【Ｏperation

Ｃompact

ＭanualdigitalＭＯＤＥＬ：ＳＰ－２４３

】

meter

Ｓeries(Analog input type)

Series nameOutput

Input

Color

Function
Preset output two points(NPN open collector output)
DC power supply(DC24V)

ＳＰ－２４３

ＡＶ３

Analog voltage output（DC1-5V）

ＡＶ４

Analog voltage output（DC0-5V）

ＡＶ５

Analog voltage output（DC0-10V）

ＡＩ

Analog current output（DC4-20mA）
Ａ２

Analog current input（DC4-20mA）

Ａ３

Analog voltage input（DC1-5V）

Ａ４

Analog voltage input（DC0-5V）

Ａ５

Analog voltage input（DC0-10V）
No
code

Gray

K

Black

！ Caution
For professional use only or designed for use by a licensed electrician only.
！ Caution
Check if the label（model name）of the unit and your desired productspecification correspond before use.
【The ７th edition ２０ Aug. ２０２０】
＠ＳＰ－２４３ＣＥ（７）－Ｅ

Precautions
Please read this operation manual including the following precautions carefully to ensure
safe use of your meter.

Warning・・・The following cases that may cause death or serious injury.
１．Do not wire while power is supplied. There is a risk of electric shock and fire.
２．Do not touch the terminals while power is supplied. There is a risk of electric shock.
３．Do not disassemble or touch the inside of the product.
There is a risk of electric shock and fire.
４．Do not use the product in places with flammable gas or ignitable substances.
５．Prepare the emergency stop or build a fail-safe system, etc. for when aproduct is break down or abnormality operating.

Caution・・・That may cause Minor injury or Property damage.
１．Use the product at the rated range power supply voltage and load.
２．Do not use the product at the following environment.
・Where there is exposed to metal powder, dust, water, chemicals, oil, etc.
・Where there is corrosive gas.
・Outdoors or in direct sunshine.
・Where condensation occurs.
・Temperature and humidity outside the rated range.
・Where there is vibration or impact.
３．Do not let metal powder, dust, water, chemicals or oil into the product.
There is a risk of break down or fire.
４．Check periodically for defects and abnormalities.
５．If the product is break down, firing, emitting smoke, overheating, abnormal noise,
etc. turn off the power immediately and stop using it.
６．Install a switch or circuit breaker where it can be operated immediately inan emergency, Then indicate that is a shutoff device.
７．Do not place the product and wiring near noise sources.
８．If there is a possibility of invasion the lightning surges, install countermeasure parts
such as a lightning arrestor in outside.
９．It can be used almost at the same time as the power is turned on,
but requires 30 minutes of power to meet all performance requirements.
１０．When cleaning, wipe with a dry cloth. Do not use organic solvents such as
benzine, thinner and alcohol.
１１．If the waterproof packing is used in a deteriorated state, the waterproof anddustproof function will be impaired. Inspect and trade it periodically.

Contents
１．About confirmation of an attachment. and a guaranteed period ・・・１
２．Specifications・・・・・・・・・・・・・・・・・・・・・・・・・・・・・２－３
３．Mounting meter・・・・・・・・・・・・・・・・・・・・・・・・・・・・４
４．Connecting terminal boards ・・・・・・・・・・・・・・・・・・・・・５
５．Construction of input circuit / output circuit・・・・・・・・・・・・・・６－７
６．Names and functions of components on front
７．The setting menu

・・・・・・・・・・・８－９

・・・・・・・・・・・・・・・・・・・・・・・・・１０－１１

８．Initial setting values and initialization ・・・・・・・・・・・・・・・・１２
９．Content and seting the each mode ・・・・・・・・・・・・・・・・・１３－２７≪１．Operating key to the mode setting≫ ・・・・・・・・・・・・・・１３≪２．Content of the each mode and set value≫・・・・・・・・・・１４－２７Mode １「Ratemeter：Setting of the indication value in the maximum
analog input 」 ・・・・・・・・・・・・・・・・・・・・・・１４
Mode ２「Ratemeter：Setting of the least significant digit andthe decimal point」
・・・・・・・・・・・・・・・・・・・・１５
Mode ３「Ratemeter：Setting of the sampling time」
・・・・・・・・・・１５
Mode ４「Setting of the LOW cut-off rate」 ・・・・・・・・・・・・・１６
Mode ５「Setting of EXT input and measurement indication」
・・・・・１６－１７Mode ６「Totalizer：Setting of the maximum total per an hour(scaling data)」・・・・・・・・・・・・・・・・・・・・・・・１７
Mode ７「Totalizer：Setting of reset dwell time, exaggerated
indication ,the decimal point 」 ・・・・・・・・・・・・・・１８－１９Mode ８「Setting of OUT1 preset-output」 ・・・・・・・・・・・・・２０－２２Mode ９「Setting of OUT2 preset-output」・・・・・・・・・・・・・・２３Mode Ａ「Totalizer：Setting of the synchronization pulse output」
・・・２４
Mode ｂ「Analog output：Setting of measurement choice andthe output digit」（option） ・・・・・・・・・・・・・・・・・２５－２６Mode Ｃ「Analog output：Setting of maximum output indication」（option） ・・・・・・・・・・・・・・・・・・・・・・・・・２７

１０．The mode protect function

・・・・・・・・・・・・・・・・・・・・２８

１１．Calling up and modifying the offset value setting ・・・・・・・・・・２９
１２．Calling up and modifying the preset value setting ・・・・・・・・・・３０
１３．Adjusting analog input and analog output
・・・・・・・・・・・・３１-３３※(An analog output is an option.)
１４．External dimensions ・・・・・・・・・・・・・・・・・・・・・・・・３３
１５．About a noise countermeasure ・・・・・・・・・・・・・・・・・・・３４
１６．Troubleshooting ・・・・・・・・・・・・・・・・・・・・・・・・・・３５

１．About confirmation of an attachment and a guaranteed period
About confirmation of an attachment．
When you received as a product, please confirm whether it includes
the following.
（１）ＳＰ－２４３(The chosen specification) ・・・・・・・・・・・・・・・・・・１（２）Fitting for fixing the body(Attachment)

・・・・・・・・・・・・・・・・・１（３）ＳＰ－２４３ Operation manual(Attachment) ・・・・・・・・・・・・・・・１（４）Rubber packing(Gray)(Attachment)・・・・・・・・・・・・・・・・・・・・１※Choosing option K, color is black.
（５）Unit label(Attachment) ・・・・・・・・・・・・・・・・・・・・・・・・・・１
If there are the mistaking parts and the missing parts, please inform a dealer
or us.（There is a case that you don't attach by convenience.）

About a guaranteed period and a guaranteed area.
１．Guaranteed period
The period a product guarantees is 4 years from a delivered day.
２．Guaranteed area
If we trouble by responsibility in whole guaranteed period, it's repaired withoutcharge at our factory. But if a product conflicted in the following matter,
it isn't a guarantee target. Please understand.
①

Case of outside of the product specifications.

②

Case of User-conducted alterations and modifications of the unit③

Case of besides our responsibility.

④

Case of safekeeping and transportation beyond the product specification
condition.

⑤

Case of an accident.

-1-２．Specifications【 Standard specifications 】
ItemSpecifications
Operation form
Ratemeter／Totalizer
A/D conversion operation
Operation system
The resolution : About 1/7000 （To full scale input.）
LOW cut-off
Inputs within 0-29 %（selectable）of the maximum analog input are cut off.
Red LED：5 digits（upper row）Character height：7mm（range：0-99999）
Display
The Indication of Ratemeter／Totalizer is changed by DISP .
Indication change（ It is necessary to set the mode №5.）
Display
Totalize indication
Case of the display of totalize, "Ｔ" LED lights up. （Green）
LED
Measurement±０.3％±１digit for analog input（２３℃）
accuracy
Setting the indication value in the maximum analog input （selectable）
Scaling
Range：0.001-10000
０－１００００
Rate
Indication area
※When indication overflowed, “9999.9” flashes on and off
meter（Decimal point position is linked to setting）
Decimal point
Displays 1 to 4 decimal points.（selectable）
Sampling time
Rate reading averaged by 0.1-100.0 sec.（selectable）
Least significant
Real, fixed at 0, or 0/5 （selectable）
digit
Measurement±０.3％±１digit for analog input（２３℃）
accuracy
Scaling
Setting the maximum total per an hour
Indication area
０－９９９９９Decimal point
Displays 1 to 4 decimal points.（selectable）
Overflow Indication ｢99999｣ flashing, 5 digits endless, or 10 digits. （selectable）
Totalizer
Synchronization with totalizer reading （enable/disable selectable）
Synchronization pulse output：1-4 digits, output width：0.01-1.99 sec.
Synchronization（selectable）
pulse output
Signal level：NPN open collector output, rating DC30V 50mA （max.）〔Use terminals no. 3-4（OUT1） ※Do not use preset output.〕
Offset value setting can establish the indication
Offsetvalue after a reset at the reach of 0-99999. （selectable）
Reset / Hold / Inhibit / Indication change （selectable）
・When specified, input from the rear terminal board causes the
Auxiliarycurrent reading to go on "hold".
input
EXT input
・Input for more than 50 ms is on for a reset, inhibit, indication change.
〔NPN open collector input and contact input are possible.〕
Input terminal
Input terminals no. 8-9
DC4-20mA
Input impedance : 250ΩCurrent input(A2)Sensor Voltage input(A3)DC1-5V
Input impedance : 220kΩ
input
DC0-5V
Input impedance : 220kΩ
Voltage input(A4)DC0-10V
Input impedance : 220kΩ
Voltage input(A5)Temperature±１００ｐｐｍ／℃ （０～５０℃）
caracteristic-2-

Output terminal
Comparative
SystemOutput mode
1 shot time
Preset value setting
Output judgment
Output circuit
Output indication
Output reset

Output terminals no. 3-4（OUT1）, 5-4（OUT2）（”ＣＯＭ” is common.）Upper limit, lower limit (immediately) and lower limit (delay) . （selectable）

Comparison, maintenance and 1 shot.（selectable）
Up to 30 ms-2 s, 8 stages.（selectable）
The
Presetting program mode setting is also selectable.
PresetJudgment output is compared with the indication value by pre-set value.
outputTwo-points NPN open collector output, maximum rating：DC30V ５０mA max
During presetting output, the OUT1 and OUT2 LEDs（Orange）are activated.
It's reset by RST key at the front or EXT input.
The presetting output function is disabled for the specified time
Time to prohibit
interval following power ON or reset. Up to 1 s-60 s, 9 stages. （selectable）
judgment〔A lower limit (delay) doesn't function.〕
Each mode setting value and totalized value is memorized by ＦＲＡＭ（The memory number of times is within 100,000 times, About 10 year
Data backup
safekeeping.）
Change by mode protect function setting
Mode protect
It's possible to change it by 「ＯＦＦ」 setting.
function
It's impossible to change it by 「ＯＮ」 setting.
After turning on the power, more than 30 minutes.
Warm up timePower supply
ＤＣ２４Ｖ（±１０％）
Power consumption ７ＶＡ ｍａｘ
Others Temperature/
０～５０℃ ３０～８０％ＲＨ（Non-condensing）
humidity conditions
Dimensions/ weight Ｗ４８×Ｈ２４×Ｄ６４ｍｍ Approx ５０ｇ（An installation adapter isn't included.）
No codeGray
Color
OptionBlack
Material of the case ＡＢＳ
Safety class
IP66 (front)
EN61326-1
EMC
EN55011(Group1 ClassA)、EN61000-4-2、EN61000-4-3、
EN61000-4-4、EN61000-4-5、EN61000-4-6【 Option 】≪ Analog output：ＡＶ３～５／ＡＩ ≫Output terminal
Output terminals no. 11-12.
Output setting
Setting of instruction value at the maximum analog output value.
ＡＶ３
ＤＣ１－ ５Ｖ Load impedance 2kΩ or more.
Voltage output
ＡＶ４
ＤＣ０－ ５Ｖ Load impedance 2kΩ or more.
ＡＶ５
ＤＣ０－１０Ｖ Load impedance 2kΩ or more.
Current output（ＡＩ）
ＤＣ４～２０ｍＡ Load impedance 500Ω or less.
“The indication value synchronizing”“The inner measured value
Measurement choicesynchronizing” （selectable）※Ratemeter／Totalizer
Output accuracy
Within ±0.3％F.S. for indicated value.（at 23℃）
Temperature caracteristic±１００ｐｐｍ／℃ （０～５０℃）
Output response
Approx ２０ｍｓ（But, an output change is time until the 90% arrival.）
14 bits Ｄ／Ａ conversion operation １３０００ resolution.
Maximum output resolving
ability
ＡＶ３
ＤＣ１－ ５Ｖ ： １３０００ ※ＡＶ４
ＤＣ０－ ５Ｖ ： １３０００ ※ＡＶ５
ＤＣ０－１０Ｖ ： １３０００ ※ＡＩ
ＤＣ４－２０ｍＡ： １３０００ ※※Maximum output area：It's possible to output to 102.4% to the
maximum of each output.
※An analog output is outputting calculation to the indication value
shown to 7segment LED. Therefore the resolution sometimes
falls from 13000 by setting of mode b,C.
-3-

３．Mounting meter
How to mount meter
１．Cut the panel to insert the meter from the front.
２．Slide fittings for fixing from the rear to fix the body.
At this time, if the body is not secured tightly, fasten screws a little more.

・Fit the body on to a panel 1.0-4.0 mm in thickness.
Fig.１

DISP

Panel cutout dimensions and, pitch for mounting two or more meters.
Fig.２（Unit：mm）※When using by protection against dust and waterproofing in forebody (IP66),
please use rubber packing.
-4-４．Connecting terminal boards
Fig.３

Terminal boards

Terminal pitch：３．５ｍｍ（Phoenix：SMKDS1/12-3.5）
Wire：ＡＷＧ３０～１６ （ＳＱ Conversion：０．０５～１．３ｍｍ２）
Convering off：５．０ｍｍ

！
！
！

！

！
！

2-wire ( 2-wire transmission type )sensor

Fig.４

Analog 4-wire type sensor

Fig.６

Analog 3-wire type sensor

Fig.５

Always turn the power OFF before commencing any wiring work.
Please confirm the specification.
DC power supply connections
If the + and - are connected in reverse by mistake, the internal protective circuit
is activated to stop the reverse current flow. In this case, disconnect, then reconnect
correctly normal operation.
＜Caution＞ The input/output wiring scheme varies with sensor type. Please refer to theconnection diagrams（Figs. 4-6）above for wiring details, to avoid damage
to the sensor or input/output circuits.
＜Caution＞ When making a connection to the terminal board, make sure thelead wire is fully and firmly inserted.
＜Caution＞ Please tighten a screw of the terminal stand surely.
＜Caution＞＜Caution＞＜Caution＞-5-５．Construction of input/output circuit
１．Senser input：Voltage / current input

Fig.７

２．EXT input：NPN open collector input

Fig.８

-6-

３．Analog output：
Voltage output（ＡＶ３～５）

Fig.９

Current output（ＡＩ）

４．Preset output/
Synchronization pulse output（NPN open collector output）-7-

Fig.１０

Fig.１１

６．Names and functions of components on front① Display unit（Red）
A B C D E

Fig.１２⑥Presetting output "OUT1"LED（Orange）
1⑦Presetting output "OUT2"LED（Orange）

2T⑧Totalizer LED（Green）
②

③

④

⑤

①Display unit（Red）
Measurement state：A measured value is indicated.
Setting state：
A・・・・・ Mode № is indicated.
B to E・・・The present set value is indicated.
：Whole pre-set value setting indicates the present set value.
：Whole offset value setting indicates the offset value.
：It indicates “L-oFF” and “L-on” at the time of mode protectfunction setting.
：AnA,An-1-4 and the present bit value are indicated at the timeof analog input and analog output adjustments mode setting.
②Mode key

MODE

Turning on
the power state：When a power supply is supplied while is pushingdown MODE ，a test mode functions.
（A stop of the test mode function is power supply off.）▲

Measurement state：Whenis pressing for 2 sec. or more，while ispushing down MODE , mode setting is called.

：When MODE is pressing for 2 sec. or more, pre-set value
is called.
：When

DISP

is pressing for 2 sec. or more，while is pushing

down

MODE

, offset value setting is called.

Setting state：Mode № (indicator A) is switched.
（1→2→3・・・9→A→b→C→1

Rise）：OUT1,2 is switched at the time of pre-set value setting.
：An-1-4 is switched at the time of analog input andanalog output adjustments mode setting.

-8-

▲

③Shift key

Measurement state：

▲

：

▲

is used to call the mode setting.
（It's on for more than 2 seconds with

MODE .）

is used to call the mode protect function，or used to change it.
（It's on for more than 2 seconds. →The state indicates it. →It's on
for more than 8 seconds just as it is. →Change）《L-oFF ⇔ L-on》▲

Setting state：moves the flash figure when each setting, to the right.
▲：When MODE is pressing while is pushing down,
mode setting is switched, "the down movement"（Indicator A）.
（C→b→A・・・4→3→2→1→C
Descent）：The output bit values decreases in An-3 and 4 at the time ofanalog input and analog output adjustments mode setting.
④Display key

DISP

Turning on：When a power supply is supplied while is pushing DISP ，
the power stateanalog input and analog output adjustments mode setting.
（A stop of the analog input and analog output adjustments mode setting
is power supply off.）
Measurement state： DISP is used to call the offset value setting.
（It's on for more than 2 seconds with

MODE .）： DISP is used to switch the ratemeter／totalizer.
（" Mode №5" in setting is needed.）
Setting state： DISP changes the flash figure when each setting.
：The output bit values increases in An-3 and 4 at the time ofanalog input and analog output adjustments mode setting.
：While DISP is being pressed, can confirm the registered bit value
at an adjusting analog input and analog output. "An-1-2"⑤Reset key

RSTTurning on：Throw power supply in withthe power state
the settings.

RST

pressed to initialize

Measurement state：Resets the totalizer without interrupting or resetting
the rate reading. It also resets the presetting output.
（When doing input (terminats №6-7), setting of" mode №5" is needed.)
Setting state： RST

is used to register and make it measurement status.

：The bit value of “An-1-4” is registered at the time ofanalog input and analog output adjustments mode setting.
⑥・⑦Presetting output LED（Orange）
Measurement：It lights up "OUT1,2" at the presetting output.
state：It lights up "OUT1" at the synchronization pulse output selection.
Setting state：The presetting output LED lights up at the time of thepresetting output value setting.
⑧Totalizer LED（Or ”×10 ＬＥＤ” ）（Green）
Measurement state：It lights up at totalizer.
：When totalizer is being "x 10", it'll change to a flash.
（" Mode №7" in setting is needed.）-9-７．The setting menu
≪The test mode function≫

Power supply OFF
Turn power supply on with
MODE being pressed.

Test mode

MODE

MODE

MODE

MODE

MODE

MODE

DISP

RST

DISP

MODE

MODE

DISP

- 10 -

≪The setting menu≫

Power supply OFF
Turn power supply on
with RST being pressed.

Initialization＜Adjusting the analog input and analog output＞

Turn power
supply on.

Turn power supply on with
DISP being pressed.

Measurement state

A n
A n _
0 1 2
A n _
4 5 6
A n _

A
1
3
2
7
3
0 4 A 8A n _ 43 b 5 8

MODE
MODE
MODE
MODE
MODE
MODE
MODE
MODE

※
※

MODE

After adjusting the setting,use RST to register it.
※While DISP is being pressed,
Power supply OFF the registered bit value can be confirmed.

＜The mode protect function＞
Press

for 2 sec. or more. Keeps pressing

for 8 sec. as it is continuously.

L _ o F F
L _ o n＜The offset value setting＞
MODE + Press DISP

for 2 sec. or more.

0 0 0 0 0
RST＜The preset value setting＞
Press MODE for 2 sec. or more.
MODE

9 9 9 9. 9
9 9 9 9. 9

MODE
RST

＜The mode setting＞
MODE + Press

MODE
MODE
MODE
MODE
MODE
MODE
MODE
MODE
MODE
MODE
MODE

for 2 sec. or more.

1. 1 0 0 0.
2. 1 03.
0 2. 0
4.
0 0
5.
0 0
6. 3. 6 0 3
7.
0 0 0
8. 0 0 0 0
9. 0 0 0 0
A. 0 0. 0 5
b.
00
C. 1 0 0 0

MODE
RST

- 11 -

MODE
MODE
MODE
MODE
MODE
MODE
MODE
MODE
MODE
MODE
MODE
MODE

８．Initial setting values and initialization
If the specifications desired by the user are requested prior to shipment,
the meter will be set these settings.
Otherwise, the regular factory settings are shown below.
Value setting of each mode
Mode № Initial setting
Notes
Ａ
Ｂ Ｃ Ｄ Ｅ Ｂ Ｃ Ｄ Ｅ
１．
２．
３．
４．
５．
６．
７．
８．
９．
Ａ．
ｂ．
Ｃ．

Table.１
Mode contents

Setting of the indication value in the maximum
analog input
Setting of the least significant digit and１ ０－ －the decimal point
０ ２. ０ －Setting of the sampling time
０ ０ － －
Setting of the LOW cut-off rate
０ ０ － －
Setting of EXT input and measurement indication
Setting of the maximum total per an hour
３. ６ ０ ３(scaling data)
Setting of reset dwell time, exaggerated
０ ０ ０ －
indication ,the decimal point
０ ０ ０ ０Setting of OUT1 preset-output
０ ０ ０ ０Setting of OUT2 preset-output
０ ０. ０ ５
Setting of the synchronization pulse output
Analog output：Setting of measurement choice００ －－（option）
and the output digit
Analog output：Setting of maximum output
１ ０ ０ ０
（option）
indication
１

０

０ ０.

Presetting output set value
Presetting
Initial settingＡ
Ｂ
Ｃ Ｄ
outputＯＵＴ１
９ ９ ９ ９.
ＯＵＴ２
９ ９ ９ ９.

Ｅ
９
９Ａ

Table.２
Notes
Ｂ Ｃ Ｄ ＥＡ

Table.３
Notes
Ｂ Ｃ Ｄ Ｅ

Offset set value
OffsetIndicationＡ
０

Initial setting
Ｂ Ｃ Ｄ
０ ０ ０

〔Initialization〕
Throw power supply in with

RST

Ｅ
０

pressed to initialize the settings.

After the initialization, the set values will be as shown in Table １, Table ２
and Table ３.
Counter data and the mode protect function are also cleared.
！ ＜Caution＞※Since an initialization changes all existing setting values to the initial
setting values, be sure to record all the setting values before an
initialization.
※In case the computer froze when unusual functioning occurred with
the normal operation, initialize according to the above procedure andset the desired value again.
- 12 -

９．Content and seting the each mode
≪1．Operating method (the mode setting)≫
When doing mode setting, please operate as follows.
Operation key▲▲

key, press

Ｃ Ｄ
１ ０
↑
０→９

Ｅ
０．

Ａ Ｂ
２．１
↑
１-Ｃ

Ｃ
０

Ｄ

Ｅ

Ａ Ｂ
Ｃ．１
↑
Ｃ-１

Ｃ
０

Ｄ
０

Ｅ
０

DISP

changes the flash figure.

Each time the key is pressed, a flash figure
is rising up. （０→１→・・・→９→０→１・・・）※In Situation, doesn't indicate by a setting
figure，up to nine.
The mode № is changed.
Each time MODE is pressed, the mode № is
rising.
〔Rise〕（１→２→・・・→Ｃ→１→２・・・）
All modes are "1-C".
When the mode № reached "C", return to"1".
The mode № is changed.
When MODE is pressing ，while is pushing
down,the mode № is descent.
〔Descent〕（Ｃ→ｂ→・・・→１→Ｃ→ｂ・・・）
When the mode № reached "1", return to
"C".
When
key is released, it'll be〔Rise〕.
▲

MODE

MODE

Ａ Ｂ Ｃ Ｄ Ｅ A figure of flash indication is shifted.
１．１→０→０→０． Each time the key is pressed, a flash figure└－－←－－┘ is shifted,to the right.

DISP

While is pushingdown

Ｅ While pushing down
０． for 2 sec. or more.

"1" appears in displays A ,the value setting
for mode №1 is shown.

Ａ Ｂ
１．１

MODE

Procedure
▲

＋

▲

MODE

IndicationＡ Ｂ Ｃ Ｄ
１．１ ０ ０▲After adjusting the setting, useregister it.

RST

to

RST

The display returns to the readings
following registration.
！ ＜Caution＞※Please make the mode protect function "L-off" at the mode setting.
If it's a condition of "L-on", it can't be changed.
About the contents of the mode protect function, please refer to,"10.The mode protect function".

- 13 -

≪２．Content of the each mode
Mode №１and set value≫

Ratemeter：Setting of the indication value in the maximum
analog inputＡ
Ｂ Ｃ Ｄ Ｅ
１． １ ０ ０ ０．
Indication value
0.001-9999（The decimal point can be set.）※When it's set as 0000, it'll be the following scaling databy correlation with the decimal point location.
0000. → 10000.
000.0 → 1000.0
00.00 → 100.00
0.000 → 10.000
Please set the indication value which is at the time of the maximum
analog input.
It's indicated at the gradient between 2 points from "0".
！ ＜Caution＞※"Mode 2. Setting of the decimal point" is the setting
which makes a decimal point indicate. Therefore it's
unrelated to the decimal point location of this setting.
Attention please.
〔Ex.〕It'll be the following setting to make it indication for "1.234"
at the maximum analog input of 10V.

4 digits and decimal point
Mode №1Ａ Ｂ Ｃ Ｄ Ｅ
１．１．２ ３ ４Mode №2

Ａ Ｂ
２．３

Ｃ
＊

Ｄ

Ｅ

- 14 -The decimal point location

Mode №２

Ratemeter：Setting of the least significant digit and the decimal pointＡ
Ｂ Ｃ Ｄ
２． １ ０Ｅ
Least significant digit
０：Real
１：Fixed at 0
２：0 or 5

Decimal point
０：0
１：0.0
２：0.00３：0.000
４：0.0000【Decimal point】
The location of the decimal point setting.（Ratemeter）【Least significant digit】
The form of indication for the least significant digit（digit on the right end）is selected.
０：Real・・・・・・・・Synchronized at the sampling time.
１：Fixed at 0 ・・・・Always, "0".
２：0 or 5・・・・・・・0-4 are expressed as 0, and 5-9 as 5.
！ ＜Caution＞※This setting is effective in avoiding flickering of leastsignificant least significant digit.
Mode №

３

Ratemeter：Setting of the sampling time
Ａ
Ｂ
３．

Ｃ Ｄ Ｅ
０ ２．０
Sampling time00.1-99.9 seconds （"00.0" is "100 sec".）【Sampling time】
Input signals are read by this time, and its average value is calculated
and indicated. Therefore, input signals are averaged and renewed by
the set time.
Use this setting for preventing flickering and for stabilizing indications.
When 0.00 sec. is set, it is 100 seconds.

- 15 -

Mode №

４

Ratemeter：Setting of the LOW cut-off rate
Ａ
Ｂ
４．

Ｃ

Ｄ Ｅ
０ ０
LOW cut-off rate
00-29（%）（"00" is a shut down.）【LOW cut-off rate】
In cases where inputs falling below a certain percentage of the current/
voltage input are not needed, the % value is inputted. Inputs of this value
or lower are disregarded without being included in rate and total readings.
〔Ex.〕When it's set the low cut rate "10%" by "0-10V" input type,does not measure by less than 1V of voltage input.

Mode №

５

Setting of EXT input and measurement indication
Ａ
Ｂ Ｃ
５．

Ｄ Ｅ
０ ０
Indicator
0：Rate meter/totalizer is switched.
1：Rate meter is fixed.
2：Totalizer is fixed.
EXT input
0：Reset1：Hold2：Inhibit
3：Indication change【Indicator】
０：Rate meter/totalizer is switched.
Rate meter/totalizer is switched in
１：Rate meter is fixed.
Rate meter is fixed and indicated.
２：Totalizer is fixed.
Totalizer is fixed and indicated.

- 16 -DISP.

【ＥＸＴ input】
The function of “terminals № 6-7” can be registered.（selectable）
0：ResetTotalizer is made offset value.
When the presetting output (OUT1,2) is output, it isreleased.
1：HoldDuring input on, "Hold" is indicated the present value.
〔Operating state : flash〕※Even the state of a hold is calculated by a computer, andthe presetting output is output by calculation.
2：Inhibit
During input on, "Inhibit" restrains sensor input.
〔Operating state : non flash〕
3：Indication change
During input on, rate meter/totalizer is switched.

！ ＜Caution＞※When choosing 1 or 2 by an indicator, an indication change
doesn't function.
Mode №

６

Totalizer：Setting of the maximum total per an hour (scaling data)Ａ
Ｂ Ｃ Ｄ Ｅ６． ３．６ ０ ３
Exponential value（10 n ）n＝ 0 - 5

Scaling data
0.01-9.99 （Fixed decimal place）（Do not specify 0.00）
The total value per an hour in the maximum analog input is specified by
inputting a 3-digit value for B-D, and the exponential value（power of 10）
for E.
The setting range is 0.01-999000.
〔Ex.〕 In case of the maximum total per an hour is "1000".
1000＝1.00 ×10 ３
↑
↑
ＢＣＤ
Ｅ

Setting will be as follows.

Ａ
Ｂ Ｃ Ｄ Ｅ６． １．０ ０ ３- 17 -

Mode №

７

Totalizer：Setting of reset dwell time, exaggerated
indication ,the decimal point
Ａ
Ｂ
７．

Ｃ
０

Ｄ
０

Ｅ
０
Decimal point
0：0
1：0.0
2：0.003：0.000
4：0.0000
Overflow indication
0：99999 flashing
1：5 digits endless
2：A place shift is carried out in the 1st
excess. 〔" T"LED flashes on and off. 〕
Furthermore, when exceeded：
99999 flashing.
Reset time
0：2 sec.
1：Immediate（at ON edge）【Reset time】
Reset time for the front Reset key is specified.
0：2 sec.
After the Reset key is pressed for 2 seconds or longer, thereading is reset.
1：Immediate
The reading is reset immediately when the Reset key is pressed.
！ ＜Caution＞※While RST and "Reset" of the EXT input are input, the presetting
output is suspended.
※"Reset" of the EXT input is always immediately.
※ RST and "Reset" of the EXT input is made offset value.
※Rate meter isn't reset.

- 18 -

【Overflow indication】
The function of “overflow indication ” can is registered.（selectable）
0：99999 flashing
Counting from 0, when the total exceeds 99999, theindication flashes.
（※ Internal totaling is continued. To resume totaling from 0, reset.）
１：5 digits endless
Display is continued endlessly. When the total exceeds
99999, the totaling is resumed from 00000.
２：Display shift to the digit to the left when the total exceeds
99999 to the 1st time.
（"T"LED flashes on and off.）
Once again when the total exceeds 99999, the indication flashes.
（Indication shifts to the left.）（※ Internal totaling is continued.

To resume totaling from 0, reset.）！ ＜Caution＞※The analog output setting is registered by "totalizer",and if thetotal exceeds 99999,the analog output value is shifted in thestate biggest (102.4%).

【Decimal point】
The location of the decimal point setting. （Totalizer）- 19 -

Mode №

８

Setting of OUT1 preset-output
Ａ
Ｂ
８． ０

Ｃ
０

Ｄ
０

Ｅ
０
Output mode（2-9：One shot output）
0：Comparison 5：100ms
1：Hold6：250ms
2：30ms7：500ms
3：50ms
8： 1sec.
4：80ms
9： 2sec.
Upper or lower limit selection
0：Upper limit
1：Lower limit（Immediate）
2：Lower limit（Delay） ！ ＜Caution＞※It's the same from
Output selection
which is chosen "1",0：Rate meter
when "1" is chosen
1：Totalizer
by"output selection".
2：Synchronization pulse output

Disabled time interval
0：０sec.
5： ８sec.
1：１sec.
6：１０sec.
2：２sec.
7：２０sec.
3：３sec.
8：３０sec.
4：６sec.
9：６０sec.
The presetting output compares display value with the preset value,
and outputs it by the result.
Refer to “12．Calling up and modifying the preset value setting “for setting the preset values.
【Disabled time interval】
The time in seconds following power startup or reset at which
the presetting output function is activated is specified.
！ ＜Caution＞※It also functions,during each setting. Attention please.
※It's the same from which is chosen "0", when "2" is chosen
by" upper or lower limit selection".

- 20 -

【Output selection】
A display to compare with a preset value is chosen.
0：Rate meter
Rate meter is compared with a preset value.
1：Totalizer
Totalizer is compared with a preset value.
2：Synchronization pulse output
It outputs synchronizing with totalizer.
In this case, a preset value is unrelated.

Refer to page 24.

【Upper and lower limits selection】
Output condition is specified.
0：Upper limit
It outputs, 「Indication value ≧ Preset value」
１：Lower limit（Immediate）
It outputs, 「Indication value ≦ Preset value」
２：Lower limit（Delay）※Only when it's Rate meter, it functions.
It outputs,「Indication value ＞ Preset value → Indication value ≦ Preset value」【Output mode】
The length of a presetting output is specified.
0：Comparison
This is output when the indication value exceeds the upper/
lower limit setting value（preset value）. When the indication
value returns to within the set range, the output is turned off.
1：HoldThis is output when the indication value exceeds the upper/
lower limit setting value（preset value）. The presetting
output,once activated, is sustained until reset.
2～9：One shot output
A pulse of pre-specified width is output once when theindication value exceeds the upper/lower limit setting value（preset value）.
“Output error：±2ms”〔Ex.〕The following settings are required in a case where thepresetting function is to be activated 6 seconds following
startup, and the presetting output is to be outputted andsustained when the reading exceeds the rate meter upper limit.
“Output selection：Rate meter”
Ａ
Ｂ
８． ４

Ｃ
０

Ｄ
０

Ｅ
１

- 21 -

【Chart at the timing of the presetting output】- 22 -

Mode №

９

Setting of OUT２ preset-output
Ａ
Ｂ
９． ０

Ｃ
０

Ｄ
０

Ｅ
０
Output mode（2-9：One shot output）
0：Comparison 5：100ms
1：Hold6：250ms◆
2：30ms7：500ms◆
3：50ms
8： 1sec.
4：80ms
9： 2sec.
◆It outputs as "one shot output" at rate meter.
It outputs as "zero return" at totalizer.
Upper or lower limit selection
0：Upper limit
1：Lower limit（Immediate）
2：Lower limit（Delay） ！ ＜Caution＞※It's the same from
Output selection
which is chosen "1",0：Rate meter
when "1" is chosen
1：Totalizer
by"output selection".

Disabled time interval
0：０sec.
5： ８sec.
1：１sec.
6：１０sec.
2：２sec.
7：２０sec.
3：３sec.
8：３０sec.
4：６sec.
9：６０sec.
Each setting is the same as that of the mode № 8“Setting of OUT1 preset-output”.
【zero return】
When the indication value exceeds the upper limit（preset value）,
this outputs one pulse of the set width one time, and the indication isreturned to the offset value and totaling is resumed.
“Output error：±2ms”
In case of "the upper limit" does function , but in case of "the lower limit"
doesn't function.
！ ＜Caution＞※The following settings are required .
「Preset value ＞ Offset value」
Always reset before starting count.

- 23 -

Mode №

Totalizer：Setting of the synchronization pulse output！ ＜Caution＞※It will function, if the totalizer synchronous pulse output is
set up by output selection of the mode №8.
“The synchronization pulse is outputted from OUT1
（terminals no. 3-4）.

ＡＡ
Ｂ Ｃ Ｄ Ｅ
Ａ． ０ ０．０ ５
Synchronization pulse output width setting
0.01-1.99 sec. （"00.0" is "0.01 sec".）
Output digit selection
0：1st digit（Ｅ）
1：2nd digit（Ｄ）
2：3rd digit（Ｃ）
3：4th digit（Ｂ）【Output digit】
The digit whose change is to trigger the pulse output is specified.
【Pulse output width】
The output width of synchronization pulse is specified.
！ ＜Caution＞※Caution is required as the synchronization output pulse isrenewed each time the output digit value changes；if thepulse width is set longer than the time interval of the output
digit change, the pulse output becomes continuous.
！ ＜Caution＞※When 0.01 sec. were set by a synchronization pulse output ,an output response is faster than indication scan,so “OUT1 LED” sometimes doesn't light up.
It's outputting.

- 24 -

Mode №

Analog output：Setting of measurement choice and the output digit※”Mode №b” is an analog output option (AV3-5/AI type).

ｂ

Ａ
Ｂ
ｂ．

Ｃ Ｄ
０

Ｅ
０
Digit selection
0：Right 4 digits：comparison（Display unitＢＣＤＥ）
1：Left 4 digits：comparison（Display unitＡＢＣＤ）
Selecting between rate/total
0：Rate meter（Synchronized with the indication value .） ◆1：Rate meter（Synchronized with the calculation value.）◆
2：Totalizer （Synchronized with the indication value .）3：Totalizer （Synchronized with the calculation value.）◆Synchronized with the sampling time.
【Selecting between rate/total】
The value to be outputted is selected.
And the indication value or
the calculation value is selected too.
0：Rate meter（Synchronized with the indication value .）It outputs to the indication value.
It outputs to the holding value,when "1" is chosen
by"mode №5 - Hold".
1：Rate meter（Synchronized with the calculation value.）
It outputs to the calculation value.
2：Totalizer （Synchronized with the indication value .）It outputs to the indication value.
It outputs to the holding value,when "1" is chosen
by"mode №5 - Hold".
3：Totalizer （Synchronized with the calculation value.）
It outputs to the calculation value.

- 25 -

【Digit selection】
It chooses "the right 4 digits" or"the left 4 digits", and setting.
Ａ

Ｂ

Ｃ

Ｄ

Ｅ

Right 4 digits
Left 4 digits！ ＜Caution＞※An analog output is outputting calculation to the indication
value shown to 7segment LED. Therefore the resolution
sometimes falls from 13000 by setting of mode №b,C.
When the analog output maximum output indication value
set "more than 1300"and "the left 4 digits", it will be 13000
resolutions,basically.
＜Addition about the analog output resolution＞
The setting of analog output sets each 4 digits of left and right.
But that's compared by at most 5 digits (left 4digits) by calculation.
The analog output is outputting to the indication value into which
the reach from a least significant digit to the highest rank
figure was divided by 13000 resolutions in relation to mode C".
(the indication value to the 1bit)〔Ex.〕The settings in a case where the analog output issynchronized with the rate reading, with a maximum output
when the reading is 10, would be as follows.

＜Condition＞
Mode №b：Digit selection→ Left 4 digits
Mode №C：Setting of maximum output indication → 0001
（Result）
Digital/analog converter is 0-13000 bits to indication value 0-10.
Therefore analog is output every 1300 bits to a change in
the indication value "1".
Therefore the resolution is 10 resolution.

- 26 -

Mode №

Analog output：Setting of maximum output indication※”Mode №C” is an analog output option (AV3-5/AI type).

ＣＡ Ｂ Ｃ Ｄ Ｅ
Ｃ．１ ０ ０ ０

Indication value
0001-9999（Do not specify 0000.）
Set an indication value of the time when the analog output ismaximum.
Set a value in four digits, neglecting the decimal point.
For example, both 500.0 and 50.00 are all right.
（It sets as "5000" in this case.）〔Ex.〕The settings in a case where the analog output issynchronized with the rate reading, with a maximum output
when the reading is 5000, would be as follows.
Ａ Ｂ
ｂ．

Ｃ
０

Ｄ

Ｅ
０
C：0〔Rate meter（Synchronized with the indication value .）〕E：0〔Digit selection（Right 4 digits）〕

Ａ Ｂ Ｃ
Ｃ．５ ０

Ｄ Ｅ
０ ０
B-E〔Setting of maximum output indication;5000〕！ ＜Caution＞※Even if the indication value goes over the indication value
setting of analog maximum output,it outputs the setting of
analog maximum output indication value.
The limited on 102.4％.
※When setting mode № d as (0000), an analog output isalways 102.4%.
※If the indication value goes over the indication value settingof analog maximum output ,the limit reaches to 102.4％.
After that,it outputs by the limited value (102.4%).
※If it's always overflow indication in spite of rate meter andtotalizer, it always outputs by the limited value (102.4%).
〔Ex.〕
When it will be making the overflow indication by setting of
the biggest indication and the left 4 digitsat at "mode № b,C",
it'll be 102.4% immediately.

- 27 -

１０．The mode protect function
When the mode protect function is made effective,mode setting.
Therefore the set value can't be changed.

DISP

operation is invalid by

In an early stage,the mode protect function is invalid .
When doing the mode protect function setting, please operate as follows.
≪Operation of the mode protect≫
Operation key

Indication

Procedure▲▲

Ａ Ｂ Ｃ Ｄ Ｅ
Press the key for 2 sec. or more.
１○
The present mode protect state is
２○ Ｌ － ｏ Ｆ Ｆ
displayed .
Ｔ○↑（The mode protect：present） 〔The regular factory setting is“L-oFF” .〕Ａ Ｂ Ｃ Ｄ Ｅ
Keep pressing
for 8 sec as it's
１○
continuously, the state of mode
２○ Ｌ － ｏ ｎprotect is changed.
Ｔ○↑（The mode protect：change）※”OFF → ON” or “ON → OFF”
It usually returns whenisstopped being pressed.

▲

▲

▲

！ ＜Caution＞※The preset value setting and the offset value setting always can bechanged.
※The mode protection function becomes “OFF”, when it's initialized.

- 28 -

１１．Calling up and modifying the offset value setting
The preset totalizer reading value to be displayed directly following a reset is
specified.
For example, if the offset value is set at "01000", the reading becomes "1000"
when reset, and the count resumes from "1000".
In order to start the count from "0", the offset value should be set as "00000".
The possible range for offset is 0-99999.
The procedure for setting the offset value is described below.

DISP

▲

▲

≪Operation of the offset value setting≫※When there are no customer requests,the initial value setting is "00000".
Procedure
Indication
Operation key
Ａ Ｂ Ｃ Ｄ Ｅ While pushing down MODE key, press DISP
１○
for 2 sec. or more.
MODE ＋ DISP
２○０ ０ ０ ０ ０ "T" LED lights up and the present offset
Ｔ●value is displayed.
Ａ Ｂ Ｃ Ｄ Ｅ Shifts the flashing indication to the digit to
the right.
１○
is pressed, shifts the
２○ ０→０→０→０→０ Each time
indication
tooneright.
Ｔ● └－－－←－－－┘
Ａ

Ｂ

Ｃ

Ｄ

Ｅ

１○
２○０
Ｔ●

１

０

０

０

Changes the value of the flashing digit.
Each time DISP is pressed, the number
goes up by one.
（0→1→・・・→9→0→・・・）
After adjusting the setting, use RST toregister it.
The display returns to the readings following
registration.

RST≪After registration≫
Ａ
RST

１○
２○
Ｔ●

Ｂ

Ｃ

Ｄ

Ｅ

１

０

０

０

The registered offset value can be displayedby pressing RST .
The totalizer count is resumed from thisvalue.

！ ＜Caution＞※The mode protection function is invalid.
（Please refer to "10. The mode protect function ".）※The decimal point is interlocked with the mode №7.
※When using "zero return" by mode №9, please be sure to set the preset value
by the following condition.
Always reset before starting count.
「Preset value ＞ Offset value」- 29 -

１２．Calling up and modifying the preset value setting
Set the preset values. "OUT1 , OUT2”.
The setting ranges are 0-99999.
The procedure for setting the preset value is described below.

▲≪Operation of the preset value setting≫
Indication
Procedure
Operation key
Ａ Ｂ Ｃ Ｄ Ｅ Press MODE for 2 sec. or more.
１●"OUT1" LED lights up and the present
MODE
２○９ ９ ９ ９. ９ preset value is displayed.
Ｔ○
Ａ Ｂ Ｃ Ｄ Ｅ Shifts the flashing indication to the digit to１●
the right.
２○９→９→９→９.→９ Each time the key is pressed, shifts theＴ○└－－－←－－－┘ indication to one right..
Ａ Ｂ Ｃ Ｄ Ｅ Changes the value of the flashing digit.
Each time DISP is pressed, the number
１●DISP
２○９ ０ ９ ９. ９ goes up by one.
（0→1→・・・→9→0→・・・）
Ｔ○

RST

Ｂ

Ｃ

Ｄ

Ｅ

９

９

９.

９

The OUT2 led lights up and the preset
value setting for OUT2 is shown.
Press
and DISP to set the desired
setting value.
▲

MODE

Ａ
１○
２●９
Ｔ○

After adjusting the setting, use RST toregister it.
The display returns to the readings following
registration.

！ ＜Caution＞※Which of the totalizer or rate meter is the preset values used by must be
selected according to the mode №8 and the mode №9.
※The decimal point is interlocked with the mode №2 for rate meter and themode №7 for totalizer.
※The mode protection function is invalid.
（Please refer to "10. The mode protect function ".）- 30 -

１３．Adjusting analog input and analog output！ ＜Caution＞※It's being adjusted according to an analog input type and an analog output
option, but when being adjusted by yourself, please setting it withthe following procedure.
When a power supply is supplied while is pressing DISP ，analog input andanalog output adjustments mode setting.
※"An-1 , 2" is an analog input adjustment.
Operation key
Indication
Procedure
Ａ Ｂ Ｃ Ｄ Ｅ When a power supply is supplied while is１○pressing DISP ，"AnA" is displayed.
DISP
２○
Ａ ｎ Ａ
Ｔ○
Ａ Ｂ Ｃ Ｄ Ｅ When MODE is pressed, "An-1" is displayed.
１○
An analog input adjustment (minimum)
MODE
２○
Ａ ｎ － １ is performed.
Ｔ○
Ａ Ｂ Ｃ Ｄ Ｅ When MODE is pressed while inputting
１○
analog minimum input of the relevant２○
０ １ ２ ３ analog type "A2-5", the bit value
MODEby the analog minimum input is indicated.
Ｔ○（The bit values）
When RST is pressed, the bit value as of
RST
it is registered as a lower limit.
「An-1,2」
While DISP is being pressed, the registered
bit value can check it.
「０ １ ２ ３．
」(It functions by "An-1,2".)
DISP

When indicating theregistered bit value,
it lights up.
Ａ

MODE

１○
２○
Ｔ○
Ａ

MODE

１○
２○
Ｔ○

Ｂ

Ｃ

Ｄ

Ｅ

Ａ

ｎ

－

２

Ｂ

Ｃ

Ｄ

Ｅ

４

５

６

７

※When indicating the registered bit value,a decimal point of a least significant digit
lights up.
When MODE is pressed, "An-2" is displayed.
An analog input adjustment (maximum)is performed.
When MODE is pressed while inputting
analog maximum input of the relevant
analog type "A2-5", the bit value
by the analog maximum input is indicated.

（The bit values）
RST

When RST is pressed, the bit value as of
it is registered as a upper limit.

- 31 -

In case of "A3(1-5V)"
Item
Voltage
Adjustment (MIN)
1.000V
Adjustment (MAX) 5.000V

In case of "A4(0-5V)"
Item
Voltage
Adjustment (MIN)
0.000V
Adjustment (MAX) 5.000V

In case of "A5(0-10V)"
Item
Voltage
Adjustment (MIN)
0.000V
Adjustment (MAX) 10.000V

In case of "A2(4-20mA)"
Item
Current
Adjustment (MIN) 4.000mA
Adjustment (MAX) 20.000mA▲※"An-3-4" is an analog output adjustment (option).
When an analog output option is not, setting of "An-3,4" isn't necessary.
Ａ Ｂ Ｃ Ｄ Ｅ When MODE is pressed, "An-3" is displayed.
１○
An analog output adjustment (minimum)
MODE
２○
Ａ ｎ － ３ is performed.
Ｔ○
Ａ Ｂ Ｃ Ｄ Ｅ Please adjust the bit value to the output
minimum value of "AV3-5,AI".
１○
２○
０ ４ Ａ ８ When DISP is pressed,the bit values
increases .
Ｔ○
MODE
When
is pressed,the bit values（The bit values）
decreases .
〔The variable range is 01EC-0764.〕
After adjusting the setting, use RST to
RSTregister the lower bit values .
Ａ Ｂ Ｃ Ｄ Ｅ When MODE is pressed, "An-4" is displayed.
１○
An analog output adjustment (maximum)
MODE
２○
Ａ ｎ － ４ is performed.
Ｔ○
Ａ Ｂ Ｃ Ｄ Ｅ Please adjust the bit value to the output
１○
maximum value of "AV3-5,AI".
２○
３ ｂ ５ ８ When DISP is pressed,the bit values
increases .
Ｔ○
MODEis pressed,the bit values（The bit values） Whendecreases .
〔The variable range is 389C-3E14.〕
After adjusting the setting, use RST to
RSTregister the upper bit values .
▲Turn theAfter registration , the power supply state

power “OFF”

is “OFF”.
Ａ

Turn thepower “ON”

Ｂ

Ｃ

Ｄ

Ｅ

１○
２○ Measurement stateＴ○

When the power supply state is “ON” once
again, a measured value indicates.

- 32 -

In case of "AV3(1-5V)"
Item
Voltage
Adjustment (MIN)
1.000V
Adjustment (MAX) 5.000V

In case of "AV4(0-5V)"
Item
Voltage
Adjustment (MIN)
0.000V
Adjustment (MAX) 5.000V

In case of "AV5(0-10V)"
Item
Voltage
Adjustment (MIN)
0.000V
Adjustment (MAX) 10.000V

In case of "AI(4-20mA)"
Item
Current
Adjustment (MIN) 4.000mA
Adjustment (MAX) 20.000mA

１４．External dimensions
External dimensions

Fig.１３

1
2
T+ -POW . IN

EXT

1 2 3 4 5 6 7 8 9 10 11 12

- +
A.IN

+A . OUT(OP)（Unit：mm）- 33 -

１５．About a noise countermeasure
When influence of noise occurred, please be careful about the following.
When doing a blackout and a malfunction by influence of noise, please beinitialized. （Refer to page 12）
Please take notes of the value setting of each modes.
If it becomes normal, please take the following measure.
And please setting it once again.
（１）Please use 3 cores of shielding wire for a sensor , separate as much as
possible from a source of noise.
（２）Please avoid a source of noise (power supply line and inverter)，make it
as short as possible.
After that,please install a sensor code.
（３）Please separate from a power supply line,in a case affected by noise.
And please install a EMI filter.
（４）The manner of the sensor cord installation.
When there is a power supply line near the sensor cord, a surge andnoise are influenced.
Therefore , install a sensor cord independently or for 50 cm or more.
Fig.１４（５）When being affected than otherequipment, please use a spark killer
like Fig.16 and take a measure.

Fig.１５

Fig.１６（６）If there is an unclear point, please even consult with use about a dealer
or us.

- 34 -

１６．Troubleshooting
When abnormality occurred, please check it as follows.
Ｎｏ．

Problem

Checking point

Solution

1

Display does not →Has it connected withthe rear terminal
appear at all.
correctly?
Is the screw tightenedcertainly?
Is the polarity of the24V DC line correct?

→Connect correctly according to"Connecting terminal boards"（Refer to page 5）.
↓
When display still does notappear, have it serviced.

2

Unusual→Check with the test
LED lighting, key
mode
switch operation,(Refer to page 10）.
preset - output,
synchronization
pulse,
analog output→Is the setting for each
Rate meterremains at "0"mode correct?
and does not↓count.
→Is the sensor input
normal?
↓↓→Is the distance of thesensor normal?
↓→Is the input systemof this meter suitable
for the output signalof the sensor?

→Initialize（Refer to page 12）.
↓When it still does not resume
normal status, have it serviced.

3

4

Indicator is
flashing“99999”.

→Check the setting again（Refer to page 14-27）.

→Check the connection of thesensor（Refer to page 5）.
Check with the test mode
(Refer to page 10）.
→The sensor lamp flash is
confirmed.
Voltage/current is measured.
→Operation manual check.
↓
When it still does not resume
normal status, have it serviced.
→Check whether the→Change the scaling data.
scaling is not too large.
（Refer to page 14 for mode №1，(Rate meter)
page 17 for mode №6 and↓page 18-19 for mode №7.）
→Overflow indication.
(Totalizer)
About a noise countermeasure↓→Refer to page 34 for→Influence of noise.
“About a noise countermeasure”.
↓
When it still does not resume
normal status, make a contact
to a dealer or us.

- 35 -

< MEMO >< MEMO >< MEMO >< MEMO >Head Office／

Technical inquiry

TEL：：+81-72--274-6001
6001

FAX：
：+81-72-274
274-6005
6005

123--1, Kami, Nishi-Ku,
Nishi Ku, Sakai-shi,
Sakai shi, Osaka, 593-8311
593 8311 JapanTokyo OfficeTEL：：+81-3-5256
5256-8311
8311

FAX：
：+81-3-5256
5256-8312
8312

URL: https://www.uinics.co.jp
Scan the QR code to access the website.

Please note that the communication costs shall be borne by the customer.

※All
All specifications may be changed without notice.