Calibration technology

Accessories
for calibration technology

Your reliable partner
As a family-run business acting globally, with 8,500 highly
qualified employees, the WIKA group of companies is aworldwide leader in pressure and temperature measurement.
The company also sets the standard in the measurementof level and flow, and in calibration technology. Founded in
1946, WIKA is today a strong and reliable partner for all therequirements of industrial measurement technology, thanks
to a broad portfolio of high-precision instruments and comprehensive services.

Calibration & service centre
WIKA offers a broad product range of calibration instruments
for the physical units of measurement for pressure andtemperature, and for electrical measurands. A multitudeof specific patents ensure unmatched performance
characteristics with many of our calibration instruments.
The range of services comprises the calibration of pressure,temperature and electrical measuring instruments in ouraccredited DKD/DAkkS calibration laboratories and a mobile
service to calibrate your instruments on site.

Contents
Pressure supply
10 bar pressure supply package (150 psi)

Model CPK-PS10

440 bar pressure supply package (580 psi)

Model CPK-PS40

6300 bar pressure supply package (4,350 psi)

Model CPK-PS300

8400 bar pressure supply package (6,000 psi)

Model CPK-PS400

10Vacuum supply package

Model CPK-VP

12

Pressure supply case

Model CPK-PC

14

Test port package, connection column, dirt trap, separator

Model CPK-TP

16

Multiple distributor

Model CPK-MD

20

AdapterModel CPK-AD

22

Model CPK-HO

24

Pressure manifold

Model CPK-PM

26

Cylinder pressure reducer

Model CPK-PR

28

Instrument adaptation

Connection components
Connecting hoses
Pressure control

Volume displacer

30

Calibration and adjustment tools
Rinsing glass

32

Filling glass

32

Pointer plier

33

Tool set for CTS movement

33

Table fixing device for fine metering valve

33

Everything at a glance

34

Calibration services

36

Calibration

38

Service products

39

WIKA worldwide

40

3

Pressure supply package,
Model CPK-PS10, Pmax = 10 bar (150 psi)
The pressure supply package CPK-PS10 is a total package ready-to-connect (plug & play) for providing pressure to WIKA
calibration instruments, adjustment benches and components. The air compressor allows working and test pressures to begenerated from atmospheric pressure to 10 bar max. (150 psi).

Applications

 Pressure generation from atmospheric pressure to 10 bar(150 psi)

 Pressure provision for pressure controllers and laboratory

measuring instruments for testing, calibrating and adjusting Working pressure supply for pressure controllers(CPC3000, CPC6000, CPC8000) for measuring ranges≤ 10 bar (150 psi)
 Test pressure supply for low-pressure manifold

Components

 10 bar (150 psi) air compressor with storage tank, filter

and pressure reducer

 Connecting hose with 6 mm Swagelok® union nut bothsides

Supplementary individual componentsConnection diagram for pressure supply package,
Model CPK-PS10, Pmax = 10 bar (150 psi)

Connecting hoses
Connection adapters
Sealing set
Replacement filters

Specifications

Pressure range

0 … 10 bar (150 psi)

Charging rate

32 l/min @ 8 bar (116 psi)Sound emission

45 dB (A)Storage volume

3.5 lConnection air outlet

G ¼ female with adapter on 6 mm Swagelok®
tube fitting

Power rating

0.34 kW

Voltage supply

AC 230 V / 50/60 Hz

Dimensions

LxWxH
340 x 300 x 340 mm (13.4 x 11.8 x 13.4 inches)

Weight21 kg (46.3 lbs.)

Featuresoil-lubricated, silent, equipped with a pressure
reducer and a 0.01 µm ultra-fine filter withmanual condensate drain

4≈ 340 mm (13.4 inches)

Air compressor

≈ 300

mm (1

1.8 inches)

10 bar air compressor (150 psi)

mm es)40 ch≈ 3 3.4 in(1

Pressure supply

10 bar (150 psi) pressure supply package, model CPK-PS10
Quick order code
Field no.

Code

VersionConnecting hose (PS10-TU)



1

2
3

Order code:

Length 1 m (3 ft), 6 mm Swagelok® union nut both sides, Pmax = 27 bar (390 psi)

Length 2 m (6 ft), 6 mm Swagelok® union nut both sides, Pmax = 27 bar (390 psi)
Length 3 m (9 ft), 6 mm Swagelok® union nut both sides, Pmax = 27 bar (390 psi)

CPK-PS10-P

- Z - Z**The order code includes a 10 bar (150 psi) air compressor and the selected connecting hose

Individual components for 10 bar (150 psi) pressure supply package, model CPK-PS10
Quick order code
Field no.

Code

VersionAccessories/individual components (PS10-A)
1

Connecting hose, length 1 m (3 ft), 6 mm Swagelok® union nut both sides, Pmax = 27 bar (390 psi)

3

Connecting hose, length 3 m (9 ft), 6 mm Swagelok® union nut both sides, Pmax = 27 bar (390 psi)

2



4

5
6

Order code:

Connecting hose, length 2 m (6 ft), 6 mm Swagelok® union nut both sides, Pmax = 27 bar (390 psi)
Connection adapter G ¼ male on 6 mm Swagelok® tube fitting

Sealing set: 5 PTFE sealing rings G ¼
Replacement ultra-fine filter 0.01µm

CPK-PS10-AZ



-Z

5

Pressure supply package,model CPK-PS40, Pmax = 40 bar (580 psi)
The pressure supply package CPK-PS40 is a total package ready-to-connect (plug & play) for providing pressure to WIKA
calibration instruments, adjustment benches and components. The air amplifier allows working and test pressures to begenerated from an air drive pressure of 2 ... 10 bar (30 ... 150 psi) at a ratio of 1 : 4 to 40 bar (580 psi) max.

Applications

 Compressed-air increase from domestic mains pressure(≤ 10 bar (150 psi) to 40 bar (580 psi) max.) Pressure provision for pressure controllers and laboratory

measuring instruments for testing, calibrating andadjusting Working pressure supply for pressure controllers(CPC3000, CPC6000, CPC8000, CPC8000-H)
 Test pressure supply for low-pressure manifolds or
adjustment benches

Components

 40 bar (580 psi) air amplifier with storage tank, filter andoutput pressure reducer mounted on a base plate Connecting hose with 6 mm Swagelok® union nut
 Pressure reducer

Supplementary individual componentsConnecting hoses
Sealing set
Noise suppression case
Filter inserts

Connection diagram for pressure supply package,
Model CPK-PS40, Pmax = 40 bar (580 psi)

Specifications

Pressure range

0 … 40 bar (580 psi)

Compressed-air drivemin. 2 bar (30 psi) /max. 10 bar (150 psi)

Transmission ratio

1:4

Sound emission

79 dB (A)Storage volume
Connection (air drive pressure)
Connection (pressure outlet)
Dimensions

5l
G ⅜ female with adapter on
6 mm Swagelok® tube fitting
G ¼ female with adapter on
6 mm Swagelok® tube fitting(L x W x H)
560 x 210 x 550 mm(22 x 8.3 x 21.7 inches)

Weight21.9 kg (48.3 lbs.)

Features

No electrical energy required
equipped with a pressure reducer
and a 5 µm filter with manual
condensate drain

6≈ 550 mm (21.7 inches)

Air amplifiers≈ 560 mm (22 inches)

40 bar (580 psi) air amplifier

Pressure supply

40 bar (580 psi) pressure supply package, model CPK-PS40
Quick order code
Field no.

Code

VersionNoise suppression case (PS40-NP)



1

Z

With

Without

Connecting hose (PS40-TU)



1

2
3

Order code:

Length 1 m (3 ft), 6 mm Swagelok® union nut both sides, Pmax = 630 bar (9,135 psi)

Length 2 m (6 ft), 6 mm Swagelok® union nut both sides, Pmax = 630 bar (9,135 psi)
Length 3 m (9 ft), 6 mm Swagelok® union nut both sides, Pmax = 630 bar (9,135 psi)

CPK-PS40-P1

- Z - Z**The order code includes a 40 bar (580 psi) air amplifier, a pressure reducer and the selected connecting hose. Optionally a noise suppression case can be selected.

Individual components for 40 bar (580 psi) pressure supply package, model CPK-PS40
Quick order code
Field no.

Code

VersionAccessories/individual components (PS40-A)
1

Connecting hose, length 1 m (3 ft), 6 mm Swagelok® union nut both sides, Pmax = 630 bar (9,135 psi)

3

Connecting hose, length 3 m (9 ft), 6 mm Swagelok® union nut both sides, Pmax = 630 bar (9,135 psi)

2
4



5

6

Connecting hose, length 2 m (6 ft), 6 mm Swagelok® union nut both sides, Pmax = 630 bar (9,135 psi)
Connecting hose for pressure reducer, length 2 m (6 ft), Pmax = 630 bar (9,135 psi)

Connection adapter G ¼ male on 6 mm Swagelok® tube fitting

Connection adapter G ½ male on 6 mm Swagelok® tube fitting

7

Sealing set: 5 Cu sealing rings each G ⅜ and G ½

9Filter insert for filter at the air inlet

8A

Order code:

Noise suppression case for installing the compressor station
Filter insert for HD filter at the air outlet

CPK-PS40-AZZZ-



-Z

7

Pressure supply package,model CPK-PS300, Pmax = 300 bar (4,350 psi)
The pressure supply package CPK-PS300 is a total package ready-to-connect (plug & play) for providing pressure to WIKA
calibration instruments, adjustment benches and components. The high-pressure compressor allows you to fill a compressedair storage cylinder from atmospheric pressure to 300 bar (4,350 psi) max. The pressure reducer assembly is used to set theworking and test pressures between 5 ... 250 bar (72.5 ... 3,625 psi) from the storage cylinder.

Applications

 Pressure generation from atmospheric pressure to

300 bar (4,350 psi)

 Pressure provision for pressure controllers and laboratory

measuring instruments for testing, calibrating andadjusting Working pressure supply for pressure controllers(CPC4000, CPC6050, CPC8000) for measuring ranges≤ 300 bar (4,350 psi)
 Test pressure supply for high-pressure manifolds or
adjustment benches

Components

 300 bar (4,350 psi) high-pressure compressor with filter

systemConnection diagram for pressure supply package,model CPK-PS300, Pmax = 250 bar (3,625 psi)

 Connecting hose with 6 mm Swagelok® union nut
 Compressed-air storage cylinder 6 litres / 300 bar(4,350 psi)

 Pressure reducer assembly (adjustable working≈ 420 mm
(16.5 inches)pressures between 5 ... 250 bar (72.5 ... 3,625 psi)) withcylinder pressure gauge, pressure controller, output
pressure gauge and shut-off valve, optionally mounted
together with the storage cylinder on a stainless steel wall
bracketSupplementary individual components
Connecting hoses
Compressed-air storage cylinder
Connection assembly
Sealing






≈ 660

mm (

26 inches)

Specifications

300 bar (4,350 psi) high-pressure compressor

High-pressure compressor

Pressure accumulator assembly (optional)

Pressure range

0 … 300 bar (4,350 psi)

Charging rate

approx. 100 l/min

Sound emission

83 dB (A)Connection air outlet

M16 x 1.5 male

Voltage supply

AV 230 Volt / 50 Hz

Power rating

2.2 kW

Dimensions

LxWxH
660 x 390 x 420 mm(26 x 15.4 x 16.5 inches)

Weight46 kg (102 lbs)

Featuresoil-lubricated, air-cooled, including filter
system for separating oil/water condensate

8

Supply pressure

Output pressure(adjustable)
permissible pressure
mediumStorage volume
WeightDimensions
Pressure connections

Pmax. 300 bar (4.350 psi)
5 … 250 bar (72 … 3,625 psi)
dry, clean air or nitrogen
6 l (1.32 gal)
14.8 kg (32.63 lbs)
LxWxH
845 x 350 x 237 mm(33.27 x 13.78 x 9.33 inches)
Inlet: 6 mm tube fitting
Outlet: 6 mm tube fitting

Pressure supply

300 bar (4,350 psi) pressure supply package, model CPK-PS300
Quick order code
Field no.

Code

VersionPressure accumulator assembly (PS300-AB)



1

Z

Compressed-air storage 300 bar (4,350 psi), 6 litres incl. wall bracket with pressure reducer assembly
Without

Connecting hose supply side (PS300-CA)

1Length 1 m (3 ft), M16 x 1.5 female, 6 mm Swagelok® union nut, Pmax = 300 bar (4,350 psi)

3

Length 3 m (9 ft), M16 x 1.5 female, 6 mm Swagelok® union nut, Pmax = 300 bar (4,350 psi)

2

Length 2 m (6 ft), M16 x 1.5 female, 6 mm Swagelok® union nut, Pmax = 300 bar (4,350 psi)

Connecting hose outlet side (PS300-CR)



1

Length 1 m (3 ft), M16 x 1.5 female, 6 mm Swagelok® union nut, Pmax = 630 bar (9,135 psi)

3

Length 3 m (9 ft), M16 x 1.5 female, 6 mm Swagelok® union nut, Pmax = 630 bar (9,135 psi)

2

Order code:

Length 2 m (6 ft), M16 x 1.5 female, 6 mm Swagelok® union nut, Pmax = 630 bar (9,135 psi)

CPK-PS300-P







- Z - Z**The order code includes a 300 bar (4,350 psi) high-pressure compressor and the selected connecting hoses. Optionally a pressure accumulator assembly can be selected.

Individual components for 250 bar (3,625 psi) pressure supply package, model CPK-PS300
Quick order code
Field no.

Code

VersionAccessories/individual components (PS300-A)
1

Connecting hose supply side, length 1 m (3 ft), M16 x 1.5 female, 6 mm Swagelok® union nut, Pmax = 300 bar (4,350 psi)

3

Connecting hose supply side, length 3 m (9 ft), M16 x 1.5 female, 6 mm Swagelok® union nut, Pmax = 300 bar (4,350 psi)

2



4

5
6
7
8
9

Order code:

Connecting hose supply side, length 2 m (6 ft), M16 x 1.5 female, 6 mm Swagelok® union nut, Pmax = 300 bar

Connecting hose outlet side, length 1 m (3 ft), M16 x 1.5 female, 6 mm Swagelok® union nut, Pmax = 630 bar (9,135 psi)
Connecting hose outlet side, length 2 m (6 ft), M16 x 1.5 female, 6 mm Swagelok® union nut, Pmax = 630 bar (9,135 psi)
Connecting hose outlet side, length 3 m (9 ft), M16 x 1.5 female, 6 mm Swagelok® union nut, Pmax = 630 bar (9,135 psi)
Compressed-air storage 300 bar (4,350 psi), 6 litres (1.32 gal)

Connection assembly including pressure reducer, pressure gauge, shut-off valve and connecting hose to the pressure accumulator
Seal for M16 x 1.5

CPK-PS300-AZZZ-



-Z

9

Pressure supply package,model CPK-PS400, Pmax = 400 bar (6,000 psi)
The pressure supply package CPK-PS400 is a total package ready-to-connect (plug & play) for providing pressure to WIKA
calibration instruments, adjustment benches and components. The compressor station allows you to generate working andtest pressures of up to 400 bar (6,000 psi) from an initial pressure of 7 ... 15 bar (102 ... 218 psi) (e.g. from a nitrogen cylinder)by means of an air drive pressure of 1 ... 6.5 bar (14.5 ... 95 psi).

Applications

 Compressed-air increase from test air (7 ... 15 bar)(102 ... 218 psi) to 400 bar (6,000 psi)

 Pressure provision for pressure controllers and laboratory

measuring instruments for testing, calibrating andadjusting Working pressure supply for pressure controllers(CPC8000) for measuring ranges ≤ 400 bar (6,000 psi)
 Test pressure supply for high-pressure manifolds or
adjustment benches

Components

 400 bar (6,000 psi) high-pressure compressor station withtubular accumulator, filter and output pressure reducer,
mounted on a stainless steel rack Connecting hose with 6 mm Swagelok® union nut
 Optional pressure reducer for test air (nitrogen) Optional screw compressor for drive air

Connection diagram for pressure supply package,
Model CPK-PS400, Pmax = 400 bar (6,000 psi)

Supplementary individual components
Connecting hoses
Silencer box
Filter insert set
Sealing set
Connection adapters

mm s)
20 che≈ 4 .5 in(16≈ 520 mm (20.5 inches)







Specifications
High-pressure compressor station
Pressure range
Compressed-air drive
Initial pressure (test air)

0 … 400 bar (6,000 psi)min. 1 bar (14.5 psi)/
max. 6.5 bar (95 psi)

7 ... 15 bar (100 ... 215 psi)

Operating medium

NitrogenTransmission ratio

1 : 15 / 1 : 75

Sound emission

79 dB (A)Storage volume

approx. 0.2 l (tubular accumulator)

Connection(air drive pressure)
Connection (test air)Connection (pressure outlet)
Dimensions
WeightFeatures10

G ½ female with adapter on
6 mm Swagelok® tube fitting
G ¼ female with adapter on
6 mm Swagelok® tube fitting
9/16-18 UNF with adapter on
6 mm Swagelok® tube fitting
LxWxH
940 x 420 x 520 mm(37 x 16.5 x 20.5 inches)
51 kg (136.6 lbs.)

No electrical energy required,equipped with a pressure reducer anda 10 µm filter≈ 940 mm (37 inches)

400 bar (6,000 psi) high-pressure compressor station

Pressure supply

400 bar (6,000 psi) pressure supply package, model CPK-PS400
Quick order code
Field no.

Code

VersionPressure supply for drive air (PS400-SU)



1

Z

Screw compressor
Without

Connecting hose outlet side (PS400-TU)



1

2
3

Length 1 m (3 ft), 6 mm Swagelok® union nut both sides, Pmax = 630 bar (9,135 psi)

Length 2 m (6 ft), 6 mm Swagelok® union nut both sides, Pmax = 630 bar (9,135 psi)
Length 3 m (9 ft), 6 mm Swagelok® union nut both sides, Pmax = 630 bar (9,135 psi)

Silencer box (PS400-NP)



1

Z

Order code:

With

Without

CPK-PS400-PZ







- Z -Z**The order code includes the high-pressure compressor station and the selected connecting hose. Optionally a screw compressor for the pressure supply of the drive air and a silencer box
can be selected. A pressure reducer for the test air is available via the model CPK-PR.

Individual components for 400 bar (6,000 psi) pressure supply package, model CPK-PS400
Quick order code
Field no.

Code

VersionAccessories/individual components (PS400-A)
1

2

Connecting hose outlet side, length 2 m (6 ft), 6 mm Swagelok® union nut both sides, Pmax = 630 bar (9,135 psi)

4

Connection adapter G ½ B male on 6 mm Swagelok® union nut

3Connecting hose outlet side, length 1 m (3 ft), 6 mm Swagelok® union nut both sides, Pmax = 630 bar (9,135 psi)

5

6
7
8
9

Order code:

Connecting hose outlet side, length 3 m (9 ft), 6 mm Swagelok® union nut both sides, Pmax = 630 bar (9,135 psi)
Connection adapter G ¼ male on 6 mm Swagelok® tube fitting
Silencer box

Filter insert set for initial pressure and high pressure and filter for drive air

Sealing set: 5 Cu sealing rings each G ¼ and G ½ and 5 sealing rings each G ¼ for pressure outlet
Connection adapter for pressure outlet 9/16-18UNF on 6 mm Swagelok® tube fitting

CPK-PS400-AZZZZ-



-Z

11

Vacuum pressure supply package,model CPK-VP
The vacuum pressure supply package CPK-VP is a total package ready-to-connect (plug & play) for providing vacuum to
WIKA calibration instruments, adjustment benches and components, consisting of an electric vacuum pump and matchingconnecting hoses. Depending on the application, you can choose between two vacuum pumps that differ with respect to theirnominal pumping speed.

Applications

 Vacuum generation
 Vacuum provision for laboratory measuring instruments,pressure manifolds and test benches for testing,calibrating and adjusting Vacuum provision for pressure controllers (CPC3000,
CPC6000, CPC8000) with absolute-pressure measuring
ranges or bidirectional measuring ranges

Components

 “Laboratory series” or “industrial series” vacuum pump
 Corrugated vacuum hose or pneumatic hose incl. adapter

Connection diagram for vacuum pressure supply package

Supplementary individual components







Corrugated vacuum hoses
Pneumatic hoses
Adapters
Clamping and centring rings
Vacuum oil
Oil mist filter insert incl. seals

Fig. left: laboratory series
Fig. right: Industrial seriesSpecifications“Laboratory series” vacuum pump
Construction
Nominal pumping speed
Final total pressure“Industrial series” vacuum pumpTwo-stage rotary vane pump, oil-lubricatedTwo-stage diaphragm pump, dry-compressing

14 m³ x h-1

1.2 m³ x h-1

5 x 10-4 mbar (7 x 10-4 psi)

8 mbar (0.116 psi)
AC 230 V, 50 Hz

AC motor
Motor power

0.45 kW0.12 kW

Connection

DN25 ISO-KF with adapter on 6 mm Swagelok® tube fitting

G ¼ female with adapter on 6 mm Swagelok® tube fitting

LxWxH
462 x 164 x 240 mm (18.2 x 6.5 x 9.4 inches)

LxWxH
312 x 154 x 207 mm (12.2 x 6.1 x 8.1 inches)

27 kg (59.5 lbs)

9.3 kg (20.5 lbs)

Dimensions in mm
Weight12

Pressure supply

Vacuum pressure supply package, model CPK-VP
Quick order code for complete vacuum pressure supply package
Field no.

Code

Package (VP-PKG)



I

L

Version“Industrial series” vacuum pump, intake volume: 1.2 m³/h, adapter G ¼ male on 6 mm Swagelok® tube fitting“Laboratory series” vacuum pump, intake volume: 15 m³/h, adapter DN25 on 6 mm Swagelok® tube fitting

Connecting hose (VP-CH)

1

Corrugated vacuum hose, length 1 m (3 ft), DN25, incl. adapter on 6 mm Swagelok®

3

Pneumatic hose, length 1 m (3 ft), 6 mm Swagelok® union nut both sides, Pmax = 27 bar (390 psi)

2
4
5

Order code:

Corrugated vacuum hose, length 1 m (3 ft), DN16, incl. adapter on 6 mm Swagelok®

Pneumatic hose, length 2 m (6 ft), 6 mm Swagelok® union nut both sides, Pmax = 27 bar (390 psi)
Pneumatic hose, length 3 m (9 ft), 6 mm Swagelok® union nut both sides, Pmax = 27 bar (390 psi)

CPK-VP-- Z - Z**The order code includes the selected vacuum pump and optionally a corrugated vacuum hose or pneumatic hose.

Individual components for vacuum pressure supply package, model CPK-VP
Quick order code for supplementary individual components for the vacuum pressure supply package
Field no.

Code

VersionAccessories/individual components (VP-A)
1

Adapter G ¼ male on 6 mm Swagelok®

3

KF25C clamping ring

2
4
5
6
7



8
9

A
B

C

D
E
F

G
H

Order code:

Adapter KF25 on 6 mm Swagelok®
KF25 centring ring

Adapter KF16 on 6 mm Swagelok®
KF16C clamping ring
KF16 centring ring

Clamping ring for union nut

Vacuum oil 2 l for “laboratory series” vacuum pump

Oil mist filter insert incl. replacement seals for “laboratory series” vacuum pump
Corrugated vacuum hose, length 1 m (3 ft), DN25
Corrugated vacuum hose, length 1 m (3 ft), DN16

Corrugated vacuum hose for extension, length 1 m (3 ft), DN25 incl. adapter
Corrugated vacuum hose for extension, length 1 m (3 ft), DN16 incl. adapter

Pneumatic hose, length 1 m (3 ft), 6 mm Swagelok® union nut both sides, Pmax = 27 bar (390 psi)
Pneumatic hose, length 2 m (6 ft), 6 mm Swagelok® union nut both sides, Pmax = 27 bar (390 psi)
Pneumatic hose, length 3 m (9 ft), 6 mm Swagelok® union nut both sides, Pmax = 27 bar (390 psi)

CPK-VP-AZ-



-Z

13

Pressure supply case,model CPK-PC
The transportable pressure supply case CPK-PC is used for on-site pressure provision. The pressure supply case is available
in a low-pressure and high-pressure version and is equipped with a 2-litre nitrogen cylinder, a precision pressure reducer with
self-venting, an inlet and output pressure gauge, overpressure protection, two test connections, fine metering valves as well asa filling hose.

Applications

 Adjustment and calibration of pressure measuring

instruments on site Pressure provision for pressure controllers (CPC3000,

CPC6000 and CPC8000)

Components

L Pressure supply case
 Filling hose

B

Pressure supply case with optional trolley

Supplementary individual componentsControls

Clamping and centring rings
Threaded inserts (see model CPK-TP)
Replacement sealing inserts
Removable trolley attachment

2

1

7

3

6

5

4
1) Storage cylinder
2) Test connections
3) Vacuum connection (DN16)
4) Fine metering valves

Specifications

5) Input / output pressure gauge
6) Pressure reducer for working pressure7) Filling connection, for filling the storage
cylinder by means of the filling hose

Pressure supply case
Pressure range-1 ... 35 bar (-15 ... 500 psi)

0 ... 200 bar (0 ... 3,000 psi)

-1 ... 200 bar (-15 ... 3,000 psi)

Working pressure control range

0 ... 35 bar (0 ... 500 psi)

1.7 ... 200 bar (25 ... 3,000 psi)

1.7 ... 200 bar (25 ... 3,000 psi)

2 litres, max. 200 bar (3,000 psi)

Pressure accumulator
Connection for vacuum pump
Test connections
Dimensions

14

DN16

-

DN16

Quick-release connector M28 x 1.5 female thread, incl. changeable threaded inserts (as standard G ½ female thread)
W x L x H : 520 x 420 x 220 mm (20.5 x 16.5 x 8.7 inches)Pressure supply

Pressure supply case, model CPK-PC
Quick order code
Field no.

Code

VersionMeasuring range (PC-MR)

1-1 ... 35 bar (-15 ... 500 psi)

3

0 ... 200 bar (0 ... 3,000 psi)

2

Order code:-1 ... 200 bar (-15 ... 3,000 psi)

CPK-PC-P- Z -Z

Individual components for pressure supply case, model CPK-PC
Quick order code
Field no.

Code

VersionAccessories/individual components (PC-A)
1

Filling hose for pressure supply case, length = 2 m (6 ft)

3Adapter for pressure supply case 6 mm Swagelok® tube fitting to DN16

2



4
5
6
7
8
9

Order code:

Adapter for pressure supply case DN25 to DN16
KF16C clamping ring
KF16 centring ring

KF25C clamping ring
KF25 centring ring

Replacement sealing sets for low-pressure metering valve
Removable trolley attachment

CPK-PC-AZ-



-Z

15

Test port package,model CPK-TP
The test port package CPK-TP is a total package ready-to-connect for adapting test items, typically, to pressure controllersof the CPC series and for reference pressure measuring instruments of the CPG series. It consists of a measuring instrument
stand or a table flange, as desired, as well as a connection column, dirt trap, separator or angled connection piece.
The test item is usually mounted without tools via a knurled nut using changeable threaded inserts.

Applications

 Adaptation of test items to pressure controllers andreference pressure measuring instruments Separator for calibrating test items with different media
 Dirt trap for avoiding impurities in the test system

Components

 Measuring instrument stand or table flange Connection column, dirt trap, separator or angled

Connection diagram test port package

connection piece Connecting hose (see model CPK-HO)

Supplementary individual components

Threaded inserts
Sealing sets
Adapter sets
Blind plugs

Specifications
Measuring instrument stand

Table flange

400 bar (6,000 psi)

2,500 bar (36,250 psi)

400 bar (6,000 psi)

1,600 bar (23,200 psi)

Pressure inlet

G ¼ female, incl. adapter on
6 mm Swagelok® tube fitting

9/16-18 UNF female

G ¼ female, incl. adapter on
6 mm Swagelok® tube fitting

9/16-18 UNF female

Pressure outlet

G ½ female

Quick-release connector M28 x 1.5 female
thread, incl. changeable threaded inserts(as standard G ½ female thread)

Pmax 1)

1) Pmax refers to a single component. In combination with other components, for example hoses, the smallest admissible pressure applies

16

G ½ female

Instrument adaptation

Quick-releaseconnector withknurled nut

112 mm
(4.4 inches)

60 mm(2.4 inches)

Threaded insert, changeable

138 mm (5.4 inches)

400 bar (6,000 psi) measuring instrument stand

2,500 bar (36,250 psi) measuring instrument stand

50 mm(2 inches)

98 mm (3.9 inches)

400 bar table flange (6,000 psi)

Connection
column2,500 bar (36,250 psi)

Pmax 1)1,000 bar (14,500 psi)

Separator withdiaphragm
1,200 bar (17,400 psi)

G ½ malePressure connection
Test item connection

Separator without
diaphragm

Dirt trap

Quick-release connector M28 x 1.5 female thread, incl. changeable threaded inserts(as standard G ½ female thread)

G ½ female thread1) Pmax refers to a single component. In combination with other components, for example hoses, the smallest admissible pressure applies

Threaded insert, changeable

219 mm (8.6 inches)

Quick-releaseconnector withknurled nut

20 mm(0.8 inch)

Quick-releaseconnector withknurled nut

84 mm (3.3 inches)
150 mm (5.9 inches)
300 mm (11.8 inches)

Threaded insert, changeable

Connection column

Dirt trap17

Test port package,model CPK-TP

Specifications
Threaded insert, changeable

133 mm (5.2 inches)

287 mm (11.3 inches)

32

Separator (with diaphragm)

Separator (without diaphragm)

Threaded insert, changeable

45° angled connection piece

18

Threaded insert,changeable

90° angled connection piece

104 mm (4.1 inches)

Quick-releaseconnector withknurled nut

151 mm (5.9 inches)

Quick-release connector
with knurled nut

Instrument adaptation
Test port package, model CPK-TP
Quick order code
Field no.

Code

VersionMeasuring instrument stand (TP-GA)1

Measuring instrument stand, Pmax = 400 bar (6,000 psi), incl. pressure connection adapter G ¼ on 6 mm Swagelok®

3

Table flange, Pmax = 400 bar (6,000 psi), incl. pressure connection adapter G ¼ on 6 mm Swagelok®

2
4

Measuring instrument stand complete with connection column, Pmax = 2,500 bar (36,250 psi), pressure connection 9/16 - 18 UNF
Table flange, Pmax = 1,600 bar (23,200 psi), pressure connection 9/16 - 18 UNF

Instrument adaptation (TP-AD)
1

Connection column 150 mm (5.9 inches) incl. threaded insert G ½ female, Pmax = 2,500 bar (36,250 psi)

4

Dirt trap -1 ... 1,000 bar (-15 ... 14,500 psi) incl. threaded insert G ½ female, 0.2 ltr.

3Connection column 84 mm (3.3 inches) incl. threaded insert G ½ female, Pmax = 2,500 bar (36,250 psi)

2

5
6

7

8

9

Order code:

Connection column 300 mm (11.8 inches) incl. threaded insert G ½ female, Pmax = 1,600 bar (23,200 psi)
Dirt trap -1 ... 1,000 bar (-15 ... 14,500 psi) incl. threaded insert G ½ female, 0.03 ltr.

Separator (without diaphragm) incl. threaded insert G ½ female, Pmax = 1,000 bar (14,500 psi)
Separator (with diaphragm) incl. threaded insert G ½ female, Pmax = 1,200 bar (17,400 psi)

Angled connection piece 90° incl. threaded insert G ½ female, Pmax = 2,500 bar (36,250 psi)
Angled connection piece 45° incl. threaded insert G ½ female, Pmax = 1,000 bar (14,500 psi)

CPK-TP-P--Z - Z - Z*

*The order code includes the selected measuring instrument stand or table flange and optionally a connection column, a dirt trap, a separator or an angled connection piece.
A connecting hose is available via the model CPK-HO.

Individual components for test port package model CPK-TP
Quick order code
Field no.

Code

VersionAccessories/individual components (TP-A)
1

Threaded insert G ½

3

Threaded insert G E (sealing insert required for short threads)

2
4
5
6
7
8
9

A



B

C

D
E
F

G
H
I

J

K
L

M
N

Threaded insert G ¼ (sealing insert required for short threads)
Threaded insert G ⅜

Threaded insert G G

Threaded insert ½ NPTThreaded insert G ¼ NPT (sealing insert required for short threads)
Threaded insert E NPT (sealing insert required for short threads)
Threaded insert ⅜ NPT (sealing insert required for short threads)

Threaded insert M10 x 1 (sealing insert required for short threads)

Threaded insert M12 x 1.5 (sealing insert required for short threads)
Threaded insert M14 x 1.5 (sealing insert required for short threads)
Threaded insert M20 x 1.5

Threaded insert M20 x 1.5 LH

Sealing insert for short threads with stud

Sealing insert for short threads without stud

O-ring set: 10 replacement seals (8 x 2) for the test connections, mat.: NBR

Sealing set for separator: 1 Viton® seal1), 2 O-rings each 10 x 2 + 24 x 3, mat.: NBR 90
Sealing set for separator: 1 butyl sealing, 2 O-rings each 10 x 2 + 24 x 3, mat.: EPDM
Sealing set: 5 replacement seals G ¼ for changeable column/dirt trap

Adapter set for quick-release connector in case G ¼, G ⅜, ½ NPT, ¼ NPT and M20 x 1.5

Adapter set “NPT” for quick-release connector in case E NPT, ¼ NPT, ⅜ NPT, and ½ NPT female
Blind plug G ½, Pmax = 2,500 bar (36,250 psi), mat.: stainless steel© 2013 WIKA Alexander Wiegand SE & Co. KG, all rights reserved.
The specifications given in this document represent the state of engineering at the time of publishing.
We reserve the right to make modifications to the specifications and materials

Order code:

CPK-TP-AZZZ--Z

1) Viton® is a registered trademark of DuPont Performance Elastomers.

19

Multiple distributor,
model CPK-MD
The multiple distributors CPK-MD are used for mounting several test items. They are available in different versions, screw-on
or stand-alone, with admissible pressure ranges between 1,600 bar (23,200 psi) and 7,000 bar (101,500 psi). The differenceis that the stand-alone version has two mounting flanges and a lateral pressure connection, allowing it to be mounted on asolid surface. The screw-on version can be adapted directly to a connection column (e.g. of a dead-weight tester) or a test port
package, using the pressure connection pointing downward.

Supplementary individual components

Applications

 Connection adapters O-ring set Threaded inserts (see model CPK-TP) Adaptation of several test items for testing, calibrating andadjusting Test benches Adaptation to connection columns, dirt traps or separators

Specifications
Multiple distributor with pressure range up to 1,600 bar (23,200 psi)Two-way distributor(screw-on)

VersionThree-way distributor(screw-on)

Five-way distributor(screw-on)

G ½ male thread

Pressure connection
lower mount

Connection location

lateralQuick-release connector M28 x 1.5 female thread, incl. changeable threaded inserts (as standard G ½ female thread)
180 mm (7.1 inches)

120 mm (4.7 inches)

180 mm (7.1 inches)

120 mm (4.7 inches)
53 mm(2.1 inches)

22 mm(0.9 inch)
35 mm(1.4 inches)

110 mm (4.3 inches)

53 mm(2.1 inches)

22 mm(0.9 inch)

35 mm(1.4 inches)

270 mm (4.7 inches)

210 mm (8.3 inches)

1,600 bar (23,200 psi) three-way distributor, screw-on

53 mm(2.1 inches)
22 mm(0.9 inch)
35 mm(1.4 inches)

498 mm (19.6 inches)

1,600 bar (23,200 psi) five-way distributor, screw-on

120 mm
(4.7 inches)

53 mm(2.1 inches)

110 mm
(4.3 inches)

1,600 bar (23,200 psi) two-way distributor, screw-on
120 mm
(4.7 inches)

110 mm (4.3 inches)

Distance of the test connections

154.5 mm (6.1 inches)

Test connections

20

Five-way distributor(stand-alone)

30 mm(1.2 inches)
71.5 mm(2.8
inches)

560 mm (22 inches)

1,600 bar (23,200 psi) five-way distributor, stand-alone

Instrument adaptation
Multiple distributor with pressure range up to 7,000 bar (101,500 psi)
VersionThree-way distributor (stand-alone)

Pressure connection

M16 x 1.5 female with sealing cone

Connection location

lateralKnurled nut G ¾ incl. changeable thread adapters with sealing cone (as standard M16 x 1.5 male thread)

Test connections

120 mm (4.7 inches)

120 mm
(4.7 inches)
107 mm
(4.2 inches)

41.75 mm(1.6 inches)

148.75 mm (5.9 inches)

Distance of the test connections

300 mm (11.8 inches)

7,000 bar (101,500 psi) three-way distributor, stand-alone

Multiple distributor, model CPK-MD
Quick order code
Field no.

CodeDistributor (MD-D)
1

VersionTwo-way distributor, screw-on, G ½ male, Pmax = 1,600 bar (23,200 psi)

2

Three-way distributor, screw-on, G ½ male, Pmax = 1,600 bar (23,200 psi)4

Five-way distributor, screw-on, G ½ male, Pmax = 1,600 bar (23,200 psi)

3

5

Order code:

Three-way distributor for high-pressure applications, stand-alone, M16 x 1.5 female, Pmax = 7,000 bar (101,500 psi)Five-way distributor, stand-alone, G ½ male, Pmax = 1,600 bar (23,200 psi)

CPK-MD-P

-Z-Z

Individual components for multiple distributor, model CPK-MD
Quick order code
Field no.

Code

VersionAccessories/individual components (MD-A)

1

O-ring set: 10 replacement seals (8 x 2) for the test connections, mat.: NBR

3

Connection adapter G ½ female on Autoclave F250/HIP HF4 (Sno-Trik), Pmax = 2,000 bar (30,000 psi)

2

Order code:

Connection adapter G ½ female on 6 mm Swagelok® tube fitting, Pmax = 400 bar (6,000 psi)

CPK-MD-AZ



-Z
21

Adapter,model CPK-AD
The adapters CPK-AD are designed for screwing them onto the hose ends of the connecting hoses of model CPK-HO.
This allows adaptation of different calibration instruments or pressure supply components. The adapters are available indifferent versions matching the pressure range.

Applications

 Adapted to connecting hoses (model CPK-HO) Connection of different calibration instruments

Versions

Depending on the pressure range, the adapters are fittedwith the following connections matching the connecting
hoses of model CPK-HO: Vacuum: small flange DN16 or DN25 (optionally also with
6 mm Swagelok® tube fitting) ≤ 630 bar (9,135 psi): 6 mm Swagelok® tube fitting ≤ 2,000 bar (30,000 psi): Autoclave F250C/HIP HF4(Sno-Trik)
The adapters are available in different thread sizes and types(see “Connection 2” in the quick order code).

Supplementary individual components

 MINIMESS® systems

22

Top: Adapter 6 mm Swagelok® on G ½ male
Bottom: Adapter 6 mm Swagelok® on G ½ female

Instrument adaptation

Adapter, model CPK-AD
Quick order code for adapters
Field no.

Code

Connection 1 (AD-C1)



1

6 mm Swagelok® tube fitting

2

Autoclave F250C/HIP HF4 (Sno-Trik)

1

G ½ (only with 6 mm Swagelok® tube fitting)3

G (only with 6 mm Swagelok® tube fitting)

5

½ NPT

7

NPT (only with 6 mm Swagelok® tube fitting)Connection 2 (AD-C2)
2

4

6



Version

G ¼ (only with 6 mm Swagelok® tube fitting)

G ⅜ (only with 6 mm Swagelok® tube fitting)

¼ NPT

8

⅜ NPT

A

Connection for pressure tapping point, MINIMESS® system, M16 x 2 G ¼ A, Pmax = 400 bar (6,000 psi) (only with 6 mm Swagelok®
tube fitting and connection type male)

9

B
C

D
E

Knurled nut M28 x 1.5 incl. threaded insert G ½ (only with female connection type)

Connection for pressure tapping point, MINIMESS® system, M16 x 1.5 G ¼ A, Pmax = 630 bar (9,135 psi) (only with 6 mm Swagelok®
tube fitting and connection type male)
M16 x 1.5 with sealing cone
M20 x 1.5 with sealing cone

9/16-18 UNF with sealing cone

Connection type (AD-CT)



F

M

Order code:

FemaleMale

CPK-AD-







-Z

Adapter, MINIMESS® systems
Description
Connection for pressure transmitter, MINIMESS® system, M16 x 2 G ½ female, Pmax = 400 bar (6,000 psi)

Connection for pressure transmitter, MINIMESS® system, M16 x 1.5 G ½ female, Pmax = 630 bar (9,135 psi)

Order number
9072306

907231423

Connecting hoses,
model CPK-HOThe connecting hoses are used for connecting calibration instruments to the pressure supply and for assembling complete
calibration set-ups. The different hose types depend on the pressure range and the medium used. They are available indifferent lengths and different connections.

Applications

 Connection between calibration instrument and pressure

supply Assembly of calibration and adjustment stations

Versions

The connecting hoses are available in different materials,
depending on the pressure range and medium.
They are available in sections up to 5 m (15 ft) max. in lengthand can be extended as desired by means of bulkhead
fittings.

Supplementary individual components

 Threaded connections for hose extension
 Connection adapter (see model CPK-AD)

Connecting hose with 6 mm Swagelok® tube fitting bothsides

Specifications
Pressure
range

Material

Allowed media

BendingradiusDiameterConnection 1)

Vacuum 2)

Stainless steel
1.4301 and 1.4404-

DN16: 146 mm
DN25: 103 mm

DN16
DN25

Small flange DN16 / DN25

≤ 27 bar (390 psi)

Polyamide

non-aggressive gases andliquids

20 mm

Inner Ø 4 mm
Outer Ø 6 mm

6 mm Swagelok® union nut≤ 630 bar (9,135 psi)

Polyamide≤ 2,000 bar(30,000 psi)

Polyoxymethylene(POM), polyamide (PA)

1)
2)

24

non-aggressive gases andliquids

20 mm

non-aggressive liquids

110 mmStandard connections on both ends, other connections are possible via adapters (see model CPK-AD)
Maximum hose length 1 m (3 ft)

Inner Ø 2 mm
Outer Ø 5 mm
Inner Ø 3.4 mm
Outer Ø 8 mm

6 mm Swagelok® union nut
Autoclave F250/HIP HF4

Connection components

Connecting hoses, model CPK-HO
Quick order code
Field no.

Code

VersionPressure range (HO-PR)
2



5
7

1

0.5 m (1.5 ft) (only possible with pressure range ≤ 2,000 bar (30,000 psi))

3

2 m (6 ft)

2
4

Medium (HO-MED)
G
H

Connection A (HO-CA)

1 m (3 ft)
3 m 9 ft

5 m (15 ft) (only possible with pressure range ≤ 2,000 bar (30.000 psi))
Gaseous

Hydraulic

1

DN25

3

Autoclave F250C/HIP HF4 (Sno-Trik)

2
4

Connection B (HO-CB)≤ 2,000 bar (30,000 psi (only possible with connections A and B: Autoclave F250C/HIP HF4 (Sno-Trik))Vacuum (only 1 m (3 ft) of hose length possible)

5≤ 630 bar (9,135 psi)V

Hose length (HO-LE)≤ 27 bar (390 psi)

6 mm Swagelok®
DN16

1

DN25

3

Autoclave F250C/HIP HF4 (Sno-Trik)

2
4

Order code:

6 mm Swagelok®
DN16

CPK-HO-P











-Z -Z

Individual components for connecting hoses, model CPK-HO
Quick order code
Field no.

Code

VersionAccessories/individual components (HO-A)

1Straight screw connection on 6 mm Swagelok® tube fitting for hose extension

3

Centring and clamping ring on DN25 for hose extension

2
4

Order code:

Straight screw connection on Sno-Trik for hose extension
Centring and clamping ring on DN16 for hose extension

CPK-HO-AZZZZZ --ZPressure manifold, model
CPK-PM
The pressure manifold CPK-PM allows adjustment and calibration of pressure measuring instruments of a wide range
of versions with gaseous media. It is available as a low-pressure and high-pressure version and equipped with two
test connections for mounting a reference instrument and test item. Fine metering valves allow the desired pressure or
vacuum to be set very precisely.

Applications

 Adjustment and calibration of pressure measuring

instruments

 Holder for test item and reference instrument
 Pressure control by means of fine metering valves

Components

The pressure manifold is available with changeable column(1) or dirt trap (2), as desired

Pressure manifold with changeable columns

Changeable column
Changeable columns are used for mounting pressure
measuring instruments.
Dirt trap
Using a dirt trap prevents impurities of the test item fromentering the test system.

Supplementary individual components

 Fine adjustment valves
 Maintenance, sealing and O-ring set Threaded inserts (see model CPK-TP)

2 1

1

150 mm(5.9 inches)2

150 mm(5.9 inches)

240 mm (9.4 inches) 240 mm (9.4 inches)
560 mm (22 inches)

Pressure manifold with changeable column (1) and dirt trap (2)Specifications
Pressure manifold
VersionPressure inlet
MediumConnection for changeable column
Test item connection

26-1 ... 30 bar (-15 ... 435 psi)

0 ... 30 bar (0 ... 435 psi)

-1 .... 400 bar (-15 ... 6,000 psi)

6 mm Swagelok® tube fitting
Clean, dry, non-corrosive gases (e.g. air or nitrogen)
G ¼ B female thread
S quick-release connector M28 x 1.5 female thread, incl. changeable threaded inserts(as standard G ½ female thread)Pressure control

Pressure manifold, model CPK-PM
Quick order code
Field no.

Code

VersionMeasuring range (PM-MR)

1-1 ... 30 bar (-15 ... 435 psi)

3

-1 ... 400 bar (-15 ... 6,000 psi)

2
4

0 ... 30 bar (0 ... 435 psi)

0 ... 400 bar (0 ... 6,000 psi)

Connection column right (PM-CCR)

1

Changeable column

3

Dirt trap -1 ... 400 bar (-15 ... 6,000 psi), large volume2

Dirt trap -1 ... 400 bar (-15 ... 6,000 psi), small volume

Connection column left (PM-CCL)

1Changeable column

3

Dirt trap -1 ... 400 bar (-15 ... 6,000 psi), large volume2

Order code:

Dirt trap -1 ... 400 bar (-15 ... 6,000 psi), small volume

CPK-PM-P- Z -Z

Individual components for pressure manifold, model CPK-PM
Quick order code
Field no.

Code

VersionAccessories/individual components (PM-A)
1

Changeable column

3

Dirt trap -1 ... 400 bar (-15 ... 6,000 psi), large volume: 0.2 litre

2
4



5
6
7
8
9

A

Order code:

Dirt trap -1 ... 400 bar (-15 ... 6,000 psi), small volume: 0.03 litre
Fine adjustment valve for 30 bar pressure manifold (435 psi)

Fine adjustment valve for 400 bar pressure manifold (6,000 psi)
Table fixing device for 30 bar fine adjustment valve (435 psi)

Maintenance set: 5 replacement seals for 30 bar fine metering valve (435 psi)
O-ring set: 10 replacement seals (8 x 2) for the test connections, mat.: NBR
Sealing set: 5 replacement seals G ¼ for changeable column/dirt trap
Blind plug G ½, Pmax = 2,500 bar (36,250 psi), mat.: stainless steel

CPK-PM-AZZZ-



-Z

27

Cylinder pressure reducer, model
CPK-PR
Precision pressure reducers CPK-PR are used for test pressure presetting in calibration and adjustment of pressure
measuring instruments. The pressure reducers are completely fitted with inlet and output pressure gauges, connection pieces
for compressed-air and nitrogen cylinders and hose connections. The pressure reducers are available in different versions,depending on the medium used, the input and output pressures and the cylinder connection.

Applications

 Pressure supply of pressure controllers (CPC3000,

CPC6000 and CPC8000)

 Test pressure presetting in adjustment and calibration

applications

Components

 Pressure reducer

One-stage pressure reducer

 Input and output pressure gauge
 Adapter on 6 mm Swagelok® tube fitting

Two-stage pressure reducer

Specifications
Cylinder pressure/input pressure 200 bar
Control range
VersionPressure outlet
Pressure inlet/cylinder connection

0 ... 170 mbar
(0 ... 2.5 psi)

0 ... 500 mbar
(0 ... 7.25 psi)

0 ... 2.1 bar(0 ... 30.5 psi)

1 ... 17 bar(15 ... 247 psi)

1 ... 50 bar(15 ... 725 psi)

5 ... 200 bar(72.5 ... 3,000 psi)Two-stage with initial pressure compensation

One-stage

E NPT female thread, incl. adapter on
6 mm Swagelok® tube fitting¼ NPT female thread, incl. adapter on
6 mm Swagelok® tube fitting

Nitrogen cylinder: W24.32 x 1/14" to DIN 477, part 1, no. 10
Compressed-air cylinder: G G to DIN 477, part 1, no. 13

Cylinder pressure/input pressure 300 bar (4,350 psi)
Control range
VersionPressure outlet
Pressure inlet/cylinder connection

28

1 ... 17 bar (15 ... 247 psi)

1 ... 50 bar (15 ... 725 psi)

5 ... 300 bar (72.5 ... 4,350 psi)

One-stage¼ NPT female thread, incl. adapter on 6 mm Swagelok® tube fitting
Nitrogen cylinder: W30 x 2 (15.9/20.1) to DIN 477, part 5, no. 54
Compressed-air cylinder: W30 x 2 (16.6/19.4) to DIN 477, part 5, no. 56

Pressure control

Pressure reducer, model CPK-PR
Quick order code for pressure reducers
Field no.

Code

Medium (PR-MED)



C

N

VersionCompressed air
Nitrogen

Cylinder pressure/input pressure (PR-IP)1
2

200 bar (3,000 psi)
300 bar (4,350 psi)

Output pressure (PR-OP)
1

0 ... 170 mbar (0 ... 2.5 psi) (only possible with 200 bar cylinder pressure/input pressure)3

0 ... 2.1 mbar (0 ... 30.5 psi) (only possible with 200 bar cylinder pressure/input pressure and pressure outlet connection E NPT)

2



4
5
7
8

0 ... 500 mbar (0 ... 7.25 psi) (only possible with 200 bar cylinder pressure/input pressure and pressure outlet connection E NPT)
1 ... 17 bar (15 ... 247 psi) (only possible with pressure outlet connection ¼ NPT)
1 ... 50 bar (15 ... 725 psi) (only possible with pressure outlet connection ¼ NPT)

5 ... 200 bar (72.5 ... 3,000 psi) (only possible with 200 bar cylinder pressure/input pressure and pressure outlet connection ¼ NPT)
5 ... 300 bar (72.5 ... 4,350 psi) (only possible with 200 bar cylinder pressure/input pressure and pressure outlet connection ¼ NPT)

Pressure outlet connection (PR-COP)



1

2

Order code:¼ NPT female incl. adapter on 6 mm Swagelok® tube fitting

E NPT female incl. adapter on 6 mm Swagelok® tube fitting

CPK-PR-



-



-



-Z

29

Volume displacer
Volume displacers are designed for fine adjustment when approaching the individual measuring points during adjustment andcalibration of pressure measuring instruments.

Applications

 Adjustment and calibration of pressure measuring

instruments

 Fine adjustment of measuring points
 Mounted to pressure manifolds, pressure supply cases orconnection columns

Special features

Volume displacers are designed for use with gaseous
test media. They are fitted with a threaded connection
G ½ B male. On the measuring instrument side the volume
displacers are fitted with a knurled nut M28 x 1.5 and athreaded insert G ½.

Supplementary individual components

 Threaded inserts (see model CPK-TP)

threaded insert,changeable≈ 124.5 mm (4.9 inches)

Quick-releaseconnector
with knurled nut≈ 218.5 mm (8.6 inches)

Volume displacer -1 ... 30 bar (-15 ... 435 psi)

30

Pressure control

Specifications
Volume displacer-1 ... 30 bar (-15 ... 435 psi)

Pressure range

G ½ male thread

Pressure connection
Test connection

Quick-release connector M28 x 1.5 female thread, incl. changeable threaded inserts (as standard G ½ female thread)

Volume displacer -1 ... 30 bar (-15 ... 435 psi), sectional view

Description

Order number

Volume displacer, -1 ... 30 bar (-15 ... 435 psi)

156541931

Calibration and adjustment tools
Rinsing glass
Applications

The rinsing glass is adapted to connection adapters andchangeable columns for rinsing measuring systems toremove impurities. On the pressure side it is fitted with athreaded connection G ½ B and on the measuring instrument
side with a knurled nut M28 x 1.5 with changeable threaded
insert G ½. Pmax 10 bar (145 psi).

Order number

1564919

Rinsing glass

Filling glass
Applications

The filling glass is used for filling measuring systems withliquid testing media prior to adjustment or calibration.
The liquid test medium in the filling glass is sucked into themeasuring system by evacuation of the measuring systemand subsequent venting. On the vacuum side it is fitted witha threaded connection G ½ male and on the measuring
instrument side with a knurled nut M28 x 1.5 with changeable
threaded insert G ½.

Order number

32

2040603

Filling glass

Rinsing glass-Sectional view

Calibration and adjustment tools

Pointer plier
Applications

The pointer pliers allow the pointers on mechanical pressure
measuring instruments to be removed without electricalaccessories.

Order number for pointer plier9091823

Order number for replacement pin

1312405

Replacement pin

Pointer plier

Table fixing device for fine adjustment valves
Applications

The table fixing device is used for mounting fine adjustment
valves on table tops. Alternatively, the table fixing device canbe used for adapting the fine adjustment valve to a pressure
manifold.

Order number

2048396

Table fixing device

33

WIKA – Your global and strong partner also incalibration technology
WIKA has successfully integrated renowned manufacturersof calibration instruments into the Group. Through the combination of experience and the product range, we can offer
each customer the ideal solution for each measurement task.
WIKA offers a unique range of products in all accuracy classes,
from primary standards to simple hand-held instruments, allwith different levels of automation.

A multitude of specific patents ensure unmatchedperformance characteristics with many of our calibration
instruments. Our customers particularly appreciate thewell-proven functionality of our instruments, which resultsfrom WIKA being not only a manufacturer of calibrationequipment, but that these are also used in their own
production and within their accredited laboratories.

The DH-Budenberg brand stands, like no other,for decades of experience in the manufacture ofpressure balances. DH-Budenberg develops andmanufactures industrial pressure balances with small
measurement uncertainties. The instruments are
suitable both for portable and stationary use.

For over 40 years, Mensor has been setting new standards
through innovative products for the automation of pressure testingtasks. The pressure controllers are suitable both for sensitive use
in laboratories and manufacturing.

Behind the WIKA brand stand portable instruments forpressure and temperature, as well as current, voltage andresistance. Thanks to their robust design, they are ideallysuited for calibration or testing tasks on site.

34

Everything at a glance

Calibration technology
From individual components ...
WIKA is the ideal partner for solutions in calibration
technology, whether a single service instrument is required
quickly on site, or whether a fully automated calibration
system needs to be designed for the laboratory or production.

We are able to offer an appropriate solution for each application.
In relation to the measuring task and the measurands, thefollowing product matrix will assist you.

Portable pressure generation

Measuring components

Hand-helds, calibrators

Test pumps serve as pressure
generators for the testing of mechanicaland electronic pressure measuring
instruments through comparative
measurements. These pressure tests cantake place in the laboratory or workshop,
or on site at the measuring point.

High-accuracy pressure sensors and very
stable standard thermometers are ideal
for applications as references in industrial
laboratories. Due to their analogue ordigital interfaces, they can be connected
to existing evaluation instruments.

Our hand-held measuring instruments(process tools) offer a simple capability
for measurement or simulation of all
established measurands on site. They
can be operated with a wide variety ofpressure sensors or thermometers.

Digitally indicating precision
measuring instruments

Digital precision instruments
and controllers

Fully automated calibration
systems as complete solutions

High-accuracy digital precision measuring
instruments are ideal for applications
as reference standards in industrial
laboratories, enabling high-accuracy
calibration. They feature exceptionallysimple handling and an extensive range offunctionality.

Due to their integrated controller,
these instruments offer exceptional
convenience. Typically, a fully automatedsetting of the required value can be set via
the interface.

Fully automated calibration systems arecustomer-specific, turnkey installations
which can be fitted in laboratories as wellas in the production environment. With
integrated reference instruments andcalibration software, calibration certificates can be generated and archived in a
simple and reproducible way.

... to a fully automated system

Pressuretemperature

current, voltage, resistance35

Calibration
Product quality, operational safety and cost effectiveness
relate directly to an accurate and reliable registration of
the process variables. Therefore, you should entrust thecalibration and maintenance of your measuring instruments
to a competent partner.

Since 1982, the WIKA calibration & service centre inKlingenberg has been a member of the German Calibration
Service (Deutscher Kalibrierdienst - DKD) and is accredited
to DIN EN ISO / IEC 17025. Since then we have been
actively participating in DKD working groups as well as
standardisation committees – contributing our experience totechnological progress.

Carrying out a calibration
Prior to calibration an evaluation of the calibration capabilityof the instruments is performed and, if necessary, anadjustment is made.
The calibration is carried out in accordance with the valid
directives. The results of the calibration are documented ina calibration certificate and the calibration item receives acalibration mark.

DKD/DAkkS calibration

List of single measured values
Specification of the applied reference standard
Calculation of mathematical parameters
Calculation of the expanded measurement uncertainty(as required in ISO 9001)

Factory calibration
 List of single measured values
 Specification of the applied reference standard Traceability to the national standard of the PTB is

 Traceability of the inspection equipment to the Documentation and graphic illustration in a traceable Documentation in an inspection certificate

 European co-operation for Accreditation ensures No normative or internationally agreed standards

 DKD/DAkkS calibration certificate

 Inspection certificate per DIN EN 10 204

guaranteed

certificate

worldwide acceptance36

We calibrate pressure, temperature and electrical measuring
instruments of our own manufacture, as well as those of othermanufacturers. Depending on your requirements you canselect either a traceable calibration or a factory calibration.

national standard is given, as a rule(no formal obligation)

Calibration & service centre

Calibration services
Our calibration laboratories have been calibrated forpressure and temperature for over 30 years. Since 2014,
our calibration laboratory has also been accredited forthe electrical measurands DC current, DC voltage andDC resistance.

 ISO 9001 certified
 DKD/DAkkS accredited (in accordance withDIN EN ISO/IEC 17025)

 Co-operation in the DKD/DAkkS working groups Over 60 years of experience in pressure and temperature

measurement

 Highly qualified, individually trained personnel
 Latest reference instruments with the highest accuracy

Manufacturer-independent calibration - fast and precise for ...
Pressure

Temperature -1 bar … +8,000 bar (-15 ... 116,000 psi)

 -196 °C … +1,200 °C (-321 °F ... +2,192 °F) with an accuracy of 0.003 % … 0.01 % of reading

 with an accuracy of 2 mK … 1.5 K or with various fixed
points (e.g. water, gallium, zinc, tin and aluminium)

Electrical measurands

On site (pressure and temperature) DC current from 0 mA ... 100 mA

In order to have the least possible impact on the production
process, we offer you a time-saving, on-site DAkkS calibration
throughout Germany (measurand pressure).

 u sing high-accuracy reference standards (pressurebalances) and working standards (precise electricalpressure measuring instruments)
 in accordance with the directives DIN EN 837,
DAkkS-DKD-R 6-1, EURAMET cg-3 or EURAMET cg-17

 DC voltage from 0 V ... 100 V

 DC resistance from 0 Ω ... 10 kΩ in accordance with the directives: VDI/VDE/DGQ/DKD 2622

 in calibration baths and tube furnaces using appropriate
reference thermometers in accordance with the appropriate DKD/DAkkS andEURAMET directives

 in our calibration van or on your workbench

 with a DAkkS accreditation for pressure- from -1 bar … +8,000 bar (-15 ... 116,000 psi)
- with accuracies between 0.025 % and 0.1 % of full scale
for the standard used

 Factory calibration for temperature from -55 ... +1,100 °C(-67 ... 2012 °F)

37

Service products
Service package “Basic”
The “Basic” service package immediately lets you plan your servicing evenbetter. You only have to compile your individual package from the services
available, define the service intervals and the duration - and we take care ofeverything else.

Service package “Take Care”
Long-term stability is one of the major criteria for a calibration instrument.
With our “Take Care” service package we not only offer you the best functionalsafety for your WIKA calibration instrument, but also reduce your service costsat the same time. In addition, you will enjoy further exclusive privileges on ourservice products - this ensures that you will appreciate your WIKA calibration
instrument for a long time!

Express service
Long downtimes of your measuring instruments are a thing of the past with the
WIKA express service. We calibrate and repair your instruments in minimal time
while meeting the highest quality claims.

Collect and return service
A date for servicing your calibration instrument always requires a little effort.
But there is another way: With our collect and return service we can transport
your measuring and calibration instruments safely, reliably and without risk.
Simply book the collect and return service at your convenience through ourservice team - we will organise the rest!

Rental and lend service
You calibrate measuring instruments in your processes and currently havecapacity shortages? Just get an additional calibration instrument - WIKA's rental
and lend service provides you quickly and without any great investments withthe adequate calibration instrument for your application.

SF6 service
Whether directly on site or in our WIKA calibration & service centre, you will
receive professional services for all SF6 products: calibration, quality analysis
and leak location. With the WIKA calibration vans, accredited to DIN EN ISO/IEC 17025, we can calibrate and repair your measuring devices directly on yourpremises.

38

Calibration & service centre

Service andconsulting

Hotline for calibration and repair
Information on calibration in the WIKA laboratory or
on-site calibrations is available from our CT service
team.
Monday - Friday from 6.30 a.m. to 4 p.m.
Phone +49 9372 132-5049
ctserviceteam@wika.com

Seminars
Our seminars on the topic of calibration technology let you
expand your know-how, further your practical knowledge andkeep you abreast of the latest developments. In addition tolearning about the theoretical basics, you will be carrying out
calibrations yourself, both manually and fully automated, withthe aid of tasks in the practical part.
 Calibration seminar, pressure
 Calibration seminar, temperature

Current dates can be found on our website.

Online services
Further information on our services and products
offering can also be found on our website.

Training
In addition to our calibration seminars, you also have theoption of receiving training customised to your needs.
No matter if for one person or a group, on your premises or at
WIKA - you design your training.
 Product training
 Calibration training Software training

Enquiry form for calibrations
Do you need a non-binding offer for acalibration? Simply use the following form.
We will be pleased to help you!

Hotline for calibration instruments
We would be pleased to assist and advise you in the selection of suitable solutions for extending your equipment pool.
Phone +49 9372 132-5015
ctsales@wika.com

39

WIKA worldwide
EuropeAustria
WIKA Messgerätevertrieb
Ursula Wiegand GmbH & Co. KG
Tel. +43 1 8691631
info@wika.at / www.wika.at
Benelux
WIKA Benelux
Tel. +31 475 535500
info@wika.nl / www.wika.nl
Bulgaria
WIKA Bulgaria EOOD
Tel. +359 2 82138-10
info@wika.bg / www.wika.bg
Croatia
WIKA Croatia d.o.o.
Tel. +385 1 6531-034
info@wika.hr / www.wika.hr
Denmark
WIKA Danmark A/S
Tel. +45 4581 9600
info@wika.as / www.wika.as
Finland
WIKA Finland Oy
Tel. +358 9 682492-0
info@wika.fi / www.wika.fi
France
WIKA Instruments s.a.r.l.
Tel. +33 1 787049-46
info@wika.fr / www.wika.fr

North America

Asia

Africa/Middle East

Romania
WIKA Instruments Romania S.R.L.
Tel. +40 21 4048327
info@wika.ro / www.wika.ro

Canada
WIKA Instruments Ltd.
Tel. +1 780 4637035
info@wika.ca / www.wika.ca

China
WIKA Instrumentation Suzhou Co., Ltd.
Tel. +86 512 6878 8000
info@wika.cn / www.wika.com.cn

Botswana
WIKA Instruments Botswana (Pty.) Ltd.
Tel. +267 3110013
info@wika.co.bw / wika.co.bw

RussiaAO “WIKA MERA”
Tel. +7 495-648018-0
info@wika.ru / www.wika.ru

USA
WIKA Instrument, LP
Tel. +1 770 5138200
info@wika.com / www.wika.us

India
WIKA Instruments India Pvt. Ltd.
Tel. +1800-123-101010
info@wika.co.in / www.wika.co.in

Egypt
WIKA Near East Ltd.
Tel. +20 2 240 13130
info@wika.com.eg / www.wika.com.eg

Serbia
WIKA Merna Tehnika d.o.o.
Tel. +381 11 2763722
info@wika.rs / www.wika.rs

Gayesco-WIKA USA, LP
Tel. +1 713 4750022
info@wikahouston.com
www.wika.us

Japan
WIKA Japan K. K.
Tel. +81 3 5439-6673
info@wika.co.jp / www.wika.co.jp

Namibia
WIKA Instruments Namibia Pty Ltd.
Tel. +26 4 61238811
info@wika.com.na / www.wika.com.na

Spain
Instrumentos WIKA S.A.U.
Tel. +34 933 9386-30
info@wika.es / www.wika.es

Mensor Corporation
Tel. +1 512 3964200
sales@mensor.com
www.mensor.com

Kazakhstan
TOO WIKA Kazakhstan
Tel. +7 727 225 9444
info@wika.kz / www.wika.kz

Nigeria
WIKA WEST AFRICA LIMITED
Tel. +234 17130019
info@wika.com.ng / www.wika.ng

Latin America

Korea
WIKA Korea Ltd.
Tel. +82 2 869-0505
info@wika.co.kr / www.wika.co.kr

Saudi Arabia
WIKA Saudi Arabia Llc
Tel. +966 53 555 0874
info@wika.sa / www.wika.sa

Malaysia
WIKA Instrumentation (M) Sdn. Bhd.
Tel. +60 3 5590 6666
info@wika.my / www.wika.my

South Africa
WIKA Instruments Pty. Ltd.
Tel. +27 11 62100-00
sales@wika.co.za / www.wika.co.za

Philippines
WIKA Instruments Philippines Inc.
Tel. +63 2 234-1270
info@wika.ph / www.wika.ph

United Arab Emirates
WIKA Middle East FZE
Tel. +971 4 883-9090
info@wika.ae / www.wika.ae
Australia
Australia

Switzerland
WIKA Schweiz AG
Tel. +41 41 91972-72
info@wika.ch / www.wika.ch
Türkiye
WIKA Instruments
Endüstriyel Ölçüm Cihazları Tic. Ltd. Şti.
Tel. +90 216 41590-66
info@wika.com.tr
www.wika.com.tr
UkraineTOV WIKA Prylad
Tel. +38 044 496 83 80
info@wika.ua / www.wika.ua

Germany
WIKA Alexander Wiegand SE & Co. KG United Kingdom
Tel. +49 9372 132-0
WIKA Instruments Ltd
info@wika.de / www.wika.de
Tel. +44 1737 644-008
info@wika.co.uk / www.wika.co.uk
Italy
WIKA Italia S.r.l. & C. S.a.s.
Tel. +39 02 93861-1
info@wika.it / www.wika.it
Poland
WIKA Polska spółka z ogranizoną
odpowiedzialnością sp. k.
Tel. +48 54 230110-0
info@wikapolska.pl
www.wikapolska.pl

Argentina
WIKA Argentina S.A.
Tel. +54 11 5442 0000
ventas@wika.com.ar
www.wika.com.ar
Brazil
WIKA do Brasil Ind. e Com. Ltda.
Tel. +55 15 3459-9700
vendas@wika.com.br
www.wika.com.br
Chile
WIKA Chile S.p.A.
Tel. +56 9 4279 0308
info@wika.cl / www.wika.cl
Colombia
Instrumentos WIKA Colombia S.A.S.
Tel. +57 601 7021347
info@wika.co / www.wika.co
MexicoInstrumentos WIKA Mexico S.A. de C.V.
Tel. +52 55 50205300
ventas@wika.com / www.wika.mx

WIKA Alexander Wiegand SE & Co. KG
Alexander-Wiegand-Straße 30 | 63911 Klingenberg | Germany
Tel. +49 9372 132-0 | info@wika.de | www.wika.deSingapore
WIKA Instrumentation Pte. Ltd.
Tel. +65 6844 5506
info@wika.sg / www.wika.sg
Taiwan
WIKA Instrumentation Taiwan Ltd.
Tel. +886 3 420 6052
info@wika.tw / www.wika.tw
Thailand
WIKA Instrumentation Corporation(Thailand) Co., Ltd.
Tel. +66 2 326 6876
info@wika.co.th / www.wika.co.th

Australia
WIKA Australia Pty. Ltd.
Tel. +61 2 88455222
sales@wika.com.au / www.wika.com.au
New Zealand
WIKA Instruments Limited
Tel. +64 9 8479020
info@wika.co.nz / www.wika.co.nz

Uzbekistan
WIKA Instrumentation FE LLC
Tel. +998 71 205 84 30
info@wika.uz / www.wika.uz

You can find further
information here!

14132521 04/ 2023 EN

www.wika.com