Optical Fiber Alignment

Alignment Components
Optical fiber alignment

Introduces core components that required alignment such as stage unit
and holders to use in WG and LD/PD alignment.

Alignment components
Stage unit
Manual stages alignment unit
Motorized stage alignment unit
Device unit
Holder

Guidance

Fiber holder

WG
alignment

Device holder
Adoptor

LD/PDalignment

Component
Cases:

Alignment
components

SMF■ SMF-WG-SMF

WG

SMF

Manualalignment unit
Motorizedalignment unit
Device unit
Fiber holders
Fiber holder

Device holders

Manual stage alignment unit

Adaptors
Contact
sensing meter
Accessories
for alignment

Device unit

StereomicroscopeLD■ LD-Lens-SMF

Lens tube

Device holder
Lens

SMF

Monitoring
unit
CCD
cameraMonitorLighting
UV equipment
Pump

Manual stages
alignment unit

Probe

Device holder

Alignment
systemAlignment
controller

Fiber holder

WG
alignment

Device unit

LD/PDalignment

Guidance for stage unit…P.4-011

4009

Guidance for holder………P.4-013

Device holder

Alignment Components Lineup
Optical fiber alignment

Manual stage alignment unit

Motorized stage alignment unit

Type of product Alignment unit (3-axis） Alignment unit (5-axis） Alignment unit (5-axis) symmetrically opposite Alignment unit (6-axis） Alignment unit (3-axis) upper free Alignment unit (3-axis）
Model
E2000S
E2000
E2001
E2200B
ES3700ES3001

AppearancePage

P.4-017

P.4-017

WEB

P.4-018

P.4-019

Motorized stage alignment unit

P.4-019

Device unit

Type of product Alignment unit (3-axis）
8°Alignment unit (4-axis） Alignment unit (5-axis） Alignment unit (6-axis)upper free Alignment unit (6-axis）
Model
ES3800ES4001ES5201ES6701ES6201

Guidance
WG
alignment

Fixing device unit
E1000

LD/PD
alignment
Alignment
components

AppearancePage

P.4-020

P.4-021

P.4-021

Fiber holder

Fiber holder for proximity

F260

F263

Rotation fiber holder
F264

P.4-022P.4-023

P.4-024

Fiber holder
Type of product
Model

Automatic rotation fiber holder Automatic rotation fiber holder for proximity
FS266
F265

Manualalignment unit
Motorizedalignment unit
Device unit

Fiber array holder
F267

Fiber holders
Device holders

Appearance

Adaptors
Page

P.4-024

P.4-024

P.4-025

P.4-026

FC connector holder

Ferrule holder

Fiber holder with lens

F262

F268

Preset base
E300

P.4-027

P.4-027

Fiber holder
Type of product
Model

Ｆ261

Preset style fiber holder Preset style V groove holder
E310
E330

Contact
sensing meter
Accessories
for alignment
Stereomicroscope
Lens tube

AppearancePage

P.4-028

Fiber holder
Type of product
Model

P.4-028

P.4-029

P.4-029

P.4-030

P.4-030

Device holder

Preset style FA holder

WG holder

WG holder

E340

F270/F271

F272/F273

Monitoring
unit
CCD
cameraMonitorPreset style WG holder Temperature control WG holder Objective lens holder
E400/E410
F274
F280/F281

Lighting
UV equipment

Appearance

Pump
Page

P.4-031

P.4-032

P.4-033

P.4-034

P.4-036

P.4-036

Probe

Device holder

AdoptorType of product Selfoc ® lens holder
Model
F290

LD holder
F125

Bracket for element adaptor
A54

Holder adaptor
F270/PB

2-axis tilt table
EB55

Tilt stage
B50/B51/B53

Alignment
systemAlignment
controller
WG
alignment

Appearance

LD/PDalignment
Page

P.4-037

P.4-037

P.4-038

P.4-038

P.4-039

P.4-039

4010

Optical Fiber Alignment

Stage Unit Guidance
Optical fiber alignment

Alignment Unit

Manual stage alignment unit (3-axis)

E2000S series

E2000 is a 3-axis unit with the high resolution manual stage (XYZ).
E2000S is sufficient if tilt alignment (θxθy) is not required.
It is the type that a tilt stage of the E2000 series became
the L-shaped bracket.

Y

X

Z（Optical axis）

Guidance
WG
alignment
LD/PDalignment

Manual stage alignment unit (5-axis)

P.4-017

E2000/
E2001 series

E2000/E2001 series is 5-axis unit withthe high resolution manual stage (XYZ) and tilt stage (θxθy) .
This is a perfect choice for the single core device alignment,
focusing manual operations.
E2001 is the symmetrically opposite XY axis of E2000.

Y
θy

θx

Z（Optical axis）

X

Alignment
components

Manual stage alignment unit (6-axis)

Manualalignment unit

P.4-017

E2200 series

E2200 series is 6-axis unit which consistsof the high resolution manual stage (XYZ), tilt stage (θxθy)and goniometer stage (θz).
This is the most efficient unit for the multi core devices,which require rotation control as fiber array or an optical waveguide.

Motorizedalignment unit
Device unit

Y
θy

θx

θz

Z（Optical axis）

X

Fiber holders
Device holders

Motorized stage alignment unit (3-axis)

Adaptors

This is combination unit of motorize XYZ + manual tilting (θxθy)
stages, and motorize stages are easy use for the single core
ES3700 series is the best choice for combining visual system due
to enough space on the above.

Contact
sensing meter

P.4-018

ES3001/ES3700 series
Y
θy

θx

Z（Optical axis）

Accessories
for alignment

X

ES3800 series is Combination unit of XYθz + manual tilting stages. P.4-020

Stereomicroscope

Motorized stage alignment unit (4-axis)

Lens tube

P.4-019

ES4001 series

ES4001 series is the unit of ES3001(Z axis) + goniometer stage (θz)
This is the most efficient unit for the multi core devices,which require rotation control as fiber array or an optical waveguide.

Monitoring
unit
CCD
camera

Y
θy

θx

θz

Z（Optical axis）

X

MonitorMotorized stage alignment unit (5-axis)

Lighting

P.4-021ES5201 seriesThis is the linier motorized XYZ stages + tilt (θxθy) stages.
This is a perfect choice for the alignment of single core devices.

UV equipment
Pump

Y
θy

θx

Z（Optical axis）

X

Probe
Alignment
systemMotorized stage alignment unit (6-axis)

Alignment
controller

ES6201/ES6701 series is the unitof linier motorized XYZ stages + tilt (θxθyθz) stages.
This is suitable for many devices, such as single mode fiber,
fiber array, optical waveguide or many optical elements,
require to align 6 directions.
ES6701 series is the best choice for combining visual system due
to enough space on the above.

WG
alignment
LD/PDalignment

4011

P.4-021

ES6201/ES6701 series
Y
θy

θz

Z（Optical axis）

General purpose alignment control system:DA200MP series：P.4-063

θxX

P.4-022

Optical fiber alignment■Moldel/Option- -ES3001

5：Y axis angular option 8A/8B
4：Select L type/R type L/R
3：Select a tilt stage
2：Overhang option W
1：Base type

Over hang plate

1.Base type

Guidance

The base type for stage unit.

WG
alignment
LD/PDalignment

180

2.Overhang option W
Attachable tilt stage in overhang to the optical axis direction with overhang plate when
it is difficult to close the device due to the interference of stage.

Center of rotation

Center of rotation

Manualalignment unit

21

21

（ See WEB for details）

Alignment
components

20

20

10

5

Motorizedalignment unit

3.Select a tilt stage
Select a tilt stage.
（

P.4-039）

A：Inner tilt

A=Inner tilt stage (B50) The center of rotation is in inside.

Device unit

B：Outer tilt

Fiber holders

B=Outer tilt stage (B51) The center of rotation is in outside.

Device holders
5

4.Select L type/R type L/R

5

Adaptors

Select a L/R of stage unit.

Contact
sensing meter
Accessories
for alignment

5.Y axis angular option 8A/8B
Used for a device end-face which is cut 8 degree.
The Y axis stage can be traveled in 8 degree tilt.

120

120

L typeR typeStereomicroscope

There are A/B type according to cutting.

Lens tube
8°

8°

8°

8°

Monitoring
unit

6.Others

CCD
cameraRemovable a fixation bracket instead of a tilt stage in E2000S.
（

See WEB for details）

MonitorCan change the stage stroke depends on model types
in ES5201/ES620 series.
（Option「10」「20」 P.4-021, 023）

120

10 10

120

10 10

120

ES3800-8A(L/R 1set Model)

120

Lighting

ES3800-8B(L/R 1set Model)

UV equipment■Correspondence table

Pump
1.Basetype

Manual stages alignment unit（3-axis） E2000S
E2000
Manual stages alignment unit（5-axis）
E2001
Manual stages alignment unit（6-axis） E2200
ES3001
Motorized stage alignment unit（3-axis） ES3700
ES3800Motorized stage alignment unit（4-axis） ES4001
Motorized stage alignment unit（5-axis） ES5201
ES6201
Motorized stage alignment unit（6-axis）
ES6701

2.Overhang
option W
W

3.Select a tilt stage
A

B

○
○
○
○

○
○
△
○

○
○
○
○

○

○

○

4.Select L type/R type
L
○
○
○
○
○
○

R
○
○
○
○
○
○

○
○
○
○

○
○
○
○

5.Y axis angular option
8Ａ
△
△
△
△

8B
△
△
△
△

○

○

△
△

△
△

6.Others

Probe

No bracket, available option

Alignment
systemAlignment
controller

L/R

1set

WG
alignment

Model

LD/PD
alignment

Available stroke change option
Available stroke change option○：Catalog standard △：Custom-designed：Without
setup

4

012

Optical Fiber Alignment

Holder Guidance
Optical fiber alignment■Fiber holder
Market of optical communications is requried high accuracy.
The holders are designed with high accuracy but easy handling for detachable.
You can select your holders based on your meeting specs.
Select a fiber from P.3-150

Guidance
WG
alignment
LD/PDalignment
Alignment
components
Manualalignment unit
Motorizedalignment unit
Device unit
Fiber holders
Device holders

Fiber holder
Fiber holder

Adaptors
Contact
sensing meter

Fiber holder for proximity

Rotation fiber holder

Automatic rotation fiber holder

Automatic rotation fiber holder for proximity

Appearance

Accessories
for alignment
Uses

Stereomicroscope

Model
Page

Lens tube
Monitoring
unit
CCD
cameraF260
P.4-024

F263
P.4-024

F264
P.4-025

F264N
P.4-025

Fiber holder
Fiber array holder

FC connector holder

Ferrule holder

Fiber holder with lensLighting
Uses

UV equipment

Model
Page

Pump

F267
P.4-027

F261
P.4-028

F262
P.4-028

F268
P.4-029

Fiber holder (preset style)
Preset base
Preset style fiber holder Preset style V groove holder Preset style FA holder

Probe
Alignment
systemAppearance

Alignment
controller
WG
alignment

Uses

LD/PD
alignment

4

013

Model
Page

FS266NP.4-026

F265
P.4-027■All holders can be mounted on
the tilt tables and stages.
All holders have key groove to match the rotating centerand optical axis. Mount on the tilt stage operates easily
with M3 screw x 2pcs.

Appearance

Monitor

FS266
P.4-026

E300
P.4-029

E310
P.4-030

E330
P.4-030

E340
P.4-031

Optical fiber alignment■Deviation of the power loss related rotating angle
Fibers are fixed by rotating holer (left-hand side) and normal holder (right-hand side) and traveling angle of the rotating holder.
(Fiber=1.3μm SMF, 10μm core diameter)
0

Measurement optical system-0.1
Power meter

LOSS

LD light source-0.2（dB）-0.3

Guidance-0.4

WGalignment

-0.5
0

50

100

150

180

The rotation angle of the holder (Deg.)

LD/PDalignment■Deviation of the time position related backlush
Fibers may distort in the part of clad and jacket. Backlush may occurs in the time of changeover the forward reserve Backlush may occurs in the time of changeover the forward reserve.
Temporal change of rotation angle may occur because twisted will return to average levels with time.
Measurement optical system
Light output

0.05
（μw）

Device unit

Leave for 2 min.
（0.5°）

Polarizing prism

MMF

0.00
725

730

735

Connection diagram
1
Blue
2
Red
3
OrangeMOTOR （PMM33BH2-C16）
4
Green
5
Black
CWLS
6
CCWLS
7
8
9
10
11
12

Motorized rotary fiber holder used
5 phase stepping motor for rotation.
Our controller DS102/DS112 arerecommended for FS266 holder driving.
Controller DS102/112: P.1-189
General purpose alignment control
system:DA200MP series: P.4-063

Device holders

740

Pulse number (Half step)Connector HR10A-10J-12P■Controller

Fiber holders

Switched
the normal orreversed

Backlash amount
（1.2°
)

PMF

Manualalignment unit
Motorizedalignment unit

0.10Power meterLD light source

Alignment
components

Adaptors
Contact
sensing meter
Accessories
for alignment
Stereomicroscope
Lens tube
Monitoring
unit
CCD
camera■Pre-set type holder

Monitor

The pre-set type holder is mounted on pre-set base.
Lighting

Quick-release with mounting a positioning pin and clamp, and also it is good for the reproducibility of the position.

UV equipment
Fiber array

Pump
Fiber array holder

Probe■Example

Alignment
systemPreset base

Alignment
controller
WG
alignment
LD/PDalignment
Tilt stage

4014

Optical Fiber Alignment

Holder Guidance
Optical fiber alignment■Device holder
There are variety of fiber device for alignment.
The device holder is designed for fixing it to the device unit and can be aligned accurately.

Guidance
WG
alignment
LD/PDalignment
Alignment
components
Manualalignment unit
Motorizedalignment unit
Device unit
Fiber holders
Device holders

Device holder
WG holder

Adaptors
Contact
sensing meter

Appearance

Accessories
for alignment
Uses

Stereomicroscope

Model
Page

Lens tube
Monitoring
unit

F270
P.4-032

F271
P.4-032

Device holder（preset style）
Preset style WG holder

CCD
cameraF272
P.4-033

F273
P.4-033

Device holder
Temperature control WG holder

Appearance

MonitorLighting

Uses

UV equipment

Model
Page

Pump

E400
P.4-034

E410
P.4-035

F274
P.4-036

Device holder

Probe

Objective lens holder

Selfoc lens ®holder

LD holder (TO-Package model)

F290
P.4-037

F125
P.4-037

Alignment
systemAppearance

Alignment
controller
WG
alignment

Uses

LD/PD
alignment

Model
Page

4

015

F280
P.4-036

F281
P.4-036

Optical fiber alignment■Mount to the device unit
The device holder is designed for mounting the device unit (E1000).
※In case of the bottom is screwed, should be removed key-plate of the device.
■Device holder application example

Guidance
WG
alignment
LD/PDalignment
Alignment
components
Manualalignment unit

Device unit

Motorizedalignment unit

■Pre-set style holder
The pre-set holder should be mounted to the pre-set base.

Device unit

It is easy to take off and on with mounting a positioning pin and clamp, and also it is good for the reproducibility of the position.

Fiber holders
Waveguide■Example

Device holders

WG holder

Adaptors
Contact
sensing meter
Accessories
for alignment
Stereomicroscope

Base※Not to be attachable to the tilt stage

Lens tube
Monitoring
unit

Device unit E1000(No key）

CCD
cameraMonitor■Holder adaptor
This adaptor is for the device holder.

Lighting

It is designed so that the optical axis height becomes 80mm by adding F270-PB10/20/26 adaptor to the device holder.
This adaptor can be helped in order to avoid any interference of the alignment.

UV equipment■Holder adaptor usage example

Pump
Probe
Alignment
systemAlignment
controller
WG
alignment
LD/PDalignment
Holder adaptor

4016

Optical Fiber Alignment
CAD
2D

Manual Stage Alignment Unit (3-axis): E2000S
Optical fiber alignment●E2000 is a 3-axis unit with the high resolution manual stage (XYZ).
●If tilt alignment (θxθy) is not required, E2000S is sufficient.

Dimensional outline drawings●It is the type that a tilt stage of the E2000 series became the L-shaped bracket.

■E2000S-L

See WEB for R type external dimension
A24-1

Y

X

Z(Optical axis)

Guidance
WG
alignment

E2000S-L

E2000S-R

Alignment
components
Manualalignment unit

109

LD/PD
alignment

165.5

SPEC
Weight
X
Y
Z
θx
θy
θz
L E2000S-L B10-60LN B30-60LK B10-60RN
̶
1.9kg
R E2000S-R B10-60RN B30-60RK B10-60LN
̶
Travel distance Coarse motion：±6.5mm Fine motion：±0.3mm
̶Resolution
Coarse motion：10μm/scale Fine：0.5μm/scale̶
Optical axis height
130mm
Accessory: Installation screws (Hexagon socket head 4 of M6-12 )■Model/Option

Device unit

６０
２５ ２５
３２

8-M4 6DP
4-M3 6DP

36

120
156

８４

L：Left type、R：Right type
Blank：Normal、O：No bracket
Unit base model number

Fiber holders

120

１６５．
５

L
E2000S 0 - R

３２
２５ ２５
６０

Motorizedalignment unit

１２０

Device holders

２８E2000S0-L

１２０
１４８

Available non-holder bracket type

Adaptors
Contact
sensing meter

CAD
2D

Manual Stage Alignment Unit (5-axis): E2000

Accessories
for alignment●E2000 series is 5-axis unit with the high resolution manual stage (XYZ) and tilt stage(θx, θy)●This is developed for the manual alignment unit, so that it is the perfect choicefor the single core device alignment.

Stereomicroscope

Dimensional outline drawings■E2000A-L

See WEB for R type external dimension
A24-1

Lens tube

YθyMonitoring
unit
CCD
cameraZ(Optical axis)

θxX

MonitorLighting

E2000A-R

SPEC
X
Y
Z
θx
θy
θz
Weight
L E2000A-L B10-60LN B30-60LK B10-60RN
B50-60LN̶
2.1kg
R E2000A-R B10-60RN B30-60RK B10-60LN
B50-60RN
̶
Travel distance Coarse motion：±6.5mm Fine motion：±0.3mm±3°̶Resolution
Coarse：10μm/scale Fine：0. 5μm/scale
33.8″/scale 33.4″/scale̶
Optical axis height
130mm
Accessory: Installation screws (Hexagon socket head 4 of M6-12 )

Pump
Probe

5
165.5

UV equipment

130

E2000A-L■Model/Option

Alignment
system

A-L
E2000 W B
R

Alignment
controller

120

LD/PD
alignment

4

017

120
155

L：Left type、R：Right type
A：Inner tilt、B：Outer tilt
Blank：Normal、W：Over hang type
Unit base model number

WG
alignment

15

120

E2000B-L（Outer tilt）

10

180

E2000WA-L（Over hang type）

Available outer chilt and over hang type

CAD
2D

Manual Stage Alignment Unit (6-axis): E2200B●This is the most effective unit for multi core device alignment, which need rotation control of optical axis such as fiber array
and an optical waveguide.

Optical fiber alignment●2200 series is 6-axis unit, which consist of the high resolution manual stage, tilt stage and goniometer stage.

Y
θy●Available an inner-tilt type. Please contact us for detail.

Z(Optical axis)

θx

θz

X

Guidance
WG
alignment
LD/PDalignment
Alignment
components

E2200B-L●Available an unit-package for splitter alignment.
●Prepare the measurement equipment (CSM) which is enabledfacing a n d G A P a d j u s t m e n t .
P.4-068 Waveguide manual alignment package（E8203B）
P.4-040 Contact sensing meter(CMS)

E2200B-R

Manualalignment unit
Motorizedalignment unit
Device unit

Dimensional outline drawings■E2200B-L

Fiber holders■E2200B-R
A24-1

Device holders

A24-1

Adaptors
Contact
sensing meter
Accessories
for alignment
Stereomicroscope
Lens tube

CCD
camera

130
165.5

165.5
130

Monitoring
unit

MonitorLighting

120

L
E2200B-LR
E2200B-R
Travel distance
Resolution
Optical axis height

10

170

170

SPEC
X
Y
Z
B10-60LN
B30-60LK
B10-60RN
B10-60RN
B30-60RK
B10-60LN
Coarse motion：±6.5mm Fine motion：±0.3mm
Coarse：10μm/scale Fine：0.5μm/scale

10

θx

θy

29.3″/scale

B51-60LN
B51-60RN±2.5°

27.8″/scale

UV equipment120θzB58-60UC
B58-60UCR±4°33″/scale

2.5kg

130mm
Accessory: Installation screws (Hexagon socket head 4 of M6-12 )■Model/Option

L
E2200 W B - R

Pump

WeightProbe
Alignment
systemAlignment
controller
WG
alignment

L：Left type、R：Right type
Blank：Normal、W：Over hang type
Unit base model number

120

LD/PD
alignment

51
171

E2200WB-L
Available over hang type

10

4

018

Optical Fiber Alignment
CAD
2D

Motorized Stage Alignment Unit (3-axis):ES3700
Dimensional outline drawings●It supports the motorized alignment of the single core device application.

■EＳ3700●ES3700 series is the best use for near focusing with a microscope and probe because

86

there is enough space on a top.

WG
alignment

ES3700-L

75

170

95

θy

θx

X

10

Z(Optical axis)

A24-5
10

Y

Guidance

See WEB for R type external dimension

150

Optical fiber alignment●This is the unit which designed for integration of the motorized stage into the XYZ axis.
Also the attitude adjustment of tilt (θxθy) can be controlled by the manual axis.

18.5

ES3700-R

123

18.5 30

Center of rotation Center of rotation
21■Example proximity probe

160
139

LD/PD
alignment
Alignment
components

236

Manualalignment unit

75

30 10

■Moldel/Option
SPEC
Weight
X
Y
Z
θx
θy
θz
L
ES3700-L EKT3-50R（※） EKS1-10L（※） EKT3-50L（※）
EB55-1L
̶
5.5kg
RES3700-R EKT3-50L（※） EKS1-10R（※） EKT3-50R（※）
EB55-1R̶
Travel distance ES3700
50mm
10mm
50mm
±2.5°
̶
1μm/Pulse(Full）0.5μm/Pulse(Full）1μm/Pulse(Full）approx.0.58°/ approx.0.57°/
̶Resolution
per rotation
0.05μm/Pulse(MS1/20） 0.05μm/Pulse(MS1/10） 0.05μm/Pulse(MS1/20） per rotation
Optical axis height
160mm
Accessory：Motor cable（D214-2-2E） Installation screws（Hexagon socket head 4 of M6-12）※Dedicated stage for alignment

Motorizedalignment unit

95

86

L
ES3700 - R

Device unit
Fiber holders
Device holders

L：Left type、R：Right type
Unit base model number

Adaptors
CAD
2D

Motorized Stage Alignment Unit (3-axis):ES3001

Contact
sensing meter●This is the unit which designed for integration of the motorized stage into the XYZ axis.
●Also the attitude adjustment of tilt (θxθy) can be controlled by the manual axis.
●It supports the motorized alignment of the single core device application.

Accessories
for alignment
Stereomicroscope
Lens tube

Dimensional outline drawings■ES3001A

See WEB for R type external dimension

YθyMonitoring
unit
CCD
cameraA24-1

Z(Optical axis)

Monitor

ES3001A-L

θx

X
R3

3

ES3001A-R
5Lighting

Pump
Probe
Alignment
system145

UV equipment

(237)
213

SPEC
Weight
X
Y
Z
θx
θy
θz
L ES3001A-L KXC06020-G KXC06020-G KXC06020-G
B50-60LN̶
2.7kg
R ES3001A-R KXC06020-G KXC06020-G KXC06020-G
B50-60RN
Travel distance
20mm
±3°
̶
1μm/PulseFull）≒33.8〃/scale ≒33.4〃/scale̶
Resolution
0.05μm/Pulse（MS1/20）
Optical axis height
145mm
Accessories：Motor cable（D214-2-2E）
Installation screws（Hexagon socket head 4 of M6-12）

20

120

■Model/Option

Alignment
controller
WG
alignment

L：Left type、R：Right type
A：Inner tilt、B：Outer tilt
Blank：Normal、W：Over hang type
Unit base model number

LD/PDalignment

120

ES3001B-L

4

019

5

10

A-L
ES3001 W BR（Outer tilt）

120
180

ES3001WA-L（over hang type）

Available outer tilt and over hang type

120
174

14

CAD
2D

Motorized Stage Alignment Unit (3-axis) Correspond to 8 degree:ES3800 Series●Also the attitude adjustment of a high resolution manual stage (θz) and tilt (θxθy) can be controlled by the manual axis.

Optical fiber alignment●This is the unit which designed for integration of the motorized stage into the linear 2-axis of XY and a rotation axis (θz).

Y●This is the most effective unit for multi core device motorized alignment,
which need rotation control of optical axis such as fiber array and an optical waveguide.
●It correspond the device that cut element edge 8 degree because Y axis stage is assembled in a 8 degree tilt position.

θy

θz

θx

8°

8°

Z(Optical axis)

X

Guidance
WG
alignment
LD/PDalignment
Alignment
components
Manualalignment unit
120

120

ES3800-8A

Motorizedalignment unit

ES3800-8B（8°）

Device unit

Dimensional outline drawings

Fiber holders■ES3800-8A（-8°）

Device holders
Adaptors
A24-1

A24-1

(R3

(R

240

Stereomicroscope

145

37
120

10

Lens tube
Monitoring
unit

145

240

(271)

3)

Accessories
for alignment

8°

)
33

(271)

8°

120

Contact
sensing meter

14

14

120
191

191

CCD
camera37
10120

MonitorLighting

L typeES3800-8A/B

L side unit
R side unit

Travel distance
Resolution

R typeX
KXC06020-G
KXC06020-G

Y
KXC06020-G
KXC06020-G
20mm

1μm/Pulse
0.05μm/Pulse（MS1/20）

Optical axis height

SPEC
Z
θx
θy
θz
Weight
B10-60RN
B51-60LN
KGW06050-L
3.5kg
B10-60LN
B51-60RN
KGW06050-R
Coarse motion：±6.5
±2.5°
±10°
Fine motion：±0.3mm
Coarse：10μm/scale≒29.3〃/scale≒27.8〃/scale
0.0045°/Full
Fine：0.5μm/scale
145mm
Accessory：Motor cable（D214-2-2E） Installation screws (Hexagon socket head 8 of M6-12)■Model/Option

UV equipment
Pump
Probe
Alignment
systemAlignment
controller
WG
alignment

A
ES3800-8 B

LD/PD
alignment

The direction of 8 degrees cutting：A、B
Unit base model number■Available the type of Y axis zero degree in conformity to a device end-face zero degree cutting.
Please contact us for details.

4020

Optical Fiber Alignment

Motorized Stage Alignment Unit (4-axis): ES4001 Series

Optical ﬁber alignment●This is the unit which designed integration of the motorized stage into
the linear 3-axis of XYZ and the rotation axis (θz) in the optical axis direction.

CAD
2D

Dimensional outline drawings

Also the attitude adjustment of tilt (θxθy) can be controlled by the manual axis.
This is the most effective unit for multi core device motorized alignment,■ES4001A-L

See WEB for R type external dimension

A24-1

which need rotation control in the optical axis direction such as fiber array and an optical waveguide.
Y
θy

θx

θz

Z(Optical axis)

X

Guidance
ES4001A-L

ES4001A-R

Motorizedalignment unit

(269)

Accessory：Motor cable（D214-2-2E)
Installation screws (Hexagon socket head 4 of M6-12)■Model/ Option

Device unit

145

Manualalignment unit

120

120

145

145

240

240

5

L：Left type、R：Right type
A：Inner tilt、B：Outer tilt
Blank：Normal、W：Over hang type
Unit base model number

Device holders

37
191

L
ES4001 W A
B-R

Fiber holders

3

3
R3

Alignment
components

R3

5

SPEC
X
Y
Z
θx
θy
θz
Weight
L ES4001A-L KXC06020-G KXC06020-G KXC06020-G
B50-60LN
KGW06050-L
3.5kg
B50-60RN
KGW06050-R
R ES4001A-R KXC06020-G KXC06020-G KXC06020-G
Travel distance
20mm
±3°
±10°
0.0045°/Pulse
1μm/ Pulse(Full)
Resolution≒33.8〃/scale≒33.4〃/scale（Full）
0.05μm/ Pulse(MS1/ 20)
Optical axis height
145mm

LD/PD
alignment240

WGalignment

Adaptors

120

10ES4001B-L

Contact
sensing meter（Outer tilt）

120

60

ES4001WA-L（over hang type）

Available outer tilt and over hang type

Accessories
for alignment

Motorized Stage Alignment Unit (5-axis): ES5201

Stereomicroscope●This is the unit which designed integration of the motorized stage into the linear 3-axis of XYZ and the tilt two axes.
●It supports the motorized alignment of the single core device application.

Lens tube

θx

Z(Optical axis)

See WEB for R type external dimension

A24-5

θy

CCD
cameraDimensional outline drawings■ES5201-LY

Monitoring
unit

CAD
2D

X

MonitorLighting
UV equipment

X
L ES5201-L KS102-30
R ES5201-R KS102-30R
L ES5211-L KS102-70
R ES5211-R KS102-70R
L ES5221-L KS102-70
R ES5221-R KS102-70R
ES5201 30mm

Pump
Probe
Alignment
systemTravel ES5211
Distance
ES5221

Alignment
controller

Resplution
Optical axis height

WG
alignment

70mm

Center of rotation

ES5201-R
SPEC
Y
Z
θx
θy
KXC06020-G KS102-70R KGW04040-L KGW06075-R
KXC06020-G KS102-70 KGW04040-R KGW06075-L
KXC06020-G KS102-70R KGW04040-L KGW06075-R
KXC06020-G KS102-70 KGW04040-R KGW06075-L
KXC06020-G KS102-100R KGW04040-L KGW06075-R
KXC06020-G KS102-100 KGW04040-R KGW06075-L
20mm

70mm
100mm

1μm/Pluse（Full）
0.05μm/Pluse（MS1/20）

θz
−
−
−
−
−
−

±8°

±8°

−

0.003°/Pluse（Full）

0.0032°/Pluse（Full）−

Weight
6.6kg
7.0kg
7.3kg

189mm

Accessory：Motor cable（D214-1-2E/ D214-2-2E）
Installation screws（Hexagon socket head 4 of M6-12）

LD/PD
alignment

4

021

189
219

ES5201-L

244.5

143

■Model/ Option

L
ES52 01
11 - R
21

L：Left type、R：Right type
Stroke type：01,11,21 ※Refer to a SPEC list
Unit base model number

214.5

14

CAD
2D

Motorized Stage Alignment Unit (6-axis): ES6701

Optical fiber alignment●This is the unit which designed for integration of the motorized stage into the linear three axes of XYZ and the tilt 3-axis (θx, θy, θz).
As this is omnidirectional controllable, so the most effective for all kinds of motorized alignment device whatever you want such asYsingle core optical fiber, fiber array, optical waveguide and each optical element.
ES6701 series is the best use for near focusing with microscope and probe because there is enough space on a top.

θy

θx

θz

Z(Optical axis)

X■Ｐroximity probe example
Guidance
WG
alignment
LD/PDalignment
Alignment
components
ES6701-L

Manualalignment unit

ES6701-R

Motorizedalignment unit

Dimensional outline drawings

Device unit

■ES6701-R

Center of rotation

10

10
95

A24-5

Device holders
Adaptors

123

63

63
2

Center of rotation

Center of rotation

Center of rotation

75
123

2

299

21

299

21

18.5

18.5

Center of rotation

18.5

10

75

150

150

Center of rotation

18.5
10

86

Fiber holders

A24-5

95

86

■ES6701-L

Contact
sensing meter
Accessories
for alignment
Stereomicroscope

76

160

63

2

75

95

86

86

95

75

Monitoring
unit

185

206

206

185

Lens tube

2

63

160

76

CCD
cameraMonitorL
ES6701-L
R
ES6701-R
Travel distance
Resolution
Optical axis height

SPEC
X
Y
Z
θx
θy
θz
WeightEKT3-50L（※）
EKS1-10L（※）
EKT3-50L（※）
KGW06100-L
KGW04040-R
KGW06075-L
6.7kg
EKT3-50R（※）
EKS1-10R（※）
EKT3-50R（※）
KGW06100-R
KGW04040-L
KGW06075-R
50mm
10mm
50mm
±6°
±8°
±8°
1μm/Pulse(Full）
0.5μm/Pulse(Full）
1μm/Pulse(Full）/Pulse（Full） 0.0030°/Pulse(Full） 0.0032°/Pulse(Full）
0.05μm/Pulse(MS1/20）0.05μm/Pulse(MS1/10）0.05μm/Pulse(MS1/20）0.002466°
206mm
Accessory：Motor cable（D214-2-2E)■Model/Option

Lighting
UV equipment
Pump
Probe

Installation screws (Hexagon socket head 4 of M6-12)※Dedicated stage for alignment

Alignment
system

L
ES6701- R
L：Left type、R：Right type

Alignment
controller
WG
alignment

Unit base model number

LD/PDalignment

4022

Optical Fiber Alignment
CAD
2D

Motorized Stage Alignment Unit (6-axis): ES6201/ES6211/ES6221
Dimensional outline drawings

As this is omnidirectional controllable, so the most effective for all kinds ofSee WEB for R type external dimension

ӖES6201-L

44.5

motorized alignment device whatever you want such as single core opticalfiber, fiber array, optical waveguide and each optical element.
ӳAvailable Y axis tilt type according to a device edge 8 degree cutting.

A24-5

10

Optical fiber alignment

ӳThis is the unit which designed for integration of the motorized stage into
the linear three axes of XYZ and the tilt 3-axis (Ġx, Ġy, Ġz).

110

Y

Center of rotationθz

Z(Optical axis)

θx

X

10

60

150

θy

18.5

Guidance

18.5
141

123
385.5

2
Center of rotation

Center of rotation
21

WGalignment

214

193

ES6211-R
244

ES6211-L

LD/PD
alignment
Alignment
components
Manualalignment unit

84.5

Motorizedalignment unit
L
R
L
R
L
R

Device unit
Fiber holders
Device holders

ES6201-L
ES6201-R
ES6211-L
ES6211-R
ES6221-L
ES6221-R
ES6201ES6211ES6221

Traveldistance

Adaptors

70mm

SPEC
Ġx
KGW04040-L
KGW04040-R
KGW04040-L
KGW04040-R
KGW04040-L
KGW04040-R

Ġy
KGW06075-R
KGW06075-L
KGW06075-R
KGW06075-L
KGW06075-R
KGW06075-L

Ġz
KGW06075-L
KGW06075-R
KGW06075-L
KGW06075-R
KGW06075-L
KGW06075-R

20mm

±8°

±8°

±8°

0.003°/Pulse
Full
214mm

0.0032°/Pulse
Full

0.0032°/Pulse
Full

Optical axis height

Accessory: Motor cableD214-1-2E/D214-2-2E)

Accessories
for alignment

2

141

Y
KXC06020-G
KXC06020-G
KXC06020-G
KXC06020-G
KXC06020-G
KXC06020-G

1μm/Pulse(Full
0.05μm/Pulse(MS1/20

Resolution

Contact
sensing meter

X
KS102-30
KS102-30R
KS102-70
KS102-70R
KS102-70
KS102-70R
30mm

160

L
ES62 01
11 - R

Lens tube

21

Monitoring
unit
CCD
camera

L：Left type、R：Right type
Stroke type：01,11,21※Refer to a SPEC list
Unit base model number

Monitor

Alignment
gnme ControllerLighting
UV equipment
Pump
Probe

SCO
ONT INUED

Alignment
systemAlignment
controllerӖӖController
mal forfo the motorized stage alignment unit.
The controller with alignment function is optimal

WG
alignment

Product Name

Model
Mod

Applicable motorized stage
Alignment unit

Number of axis

Alignment function

LD/PDalignment

Alignment control
for splitter production
ction

DA200

ES38006-axisFlat,Rotate

General purpose
urpose alignment
control system
contr

DA200MPCustomized 

ES series etc..

4-axis8-axis

One,Flat,Rotate,Focus
One,Flat,Rotate,Focus

4023

.4-063


110

44.5

Weight
7.2kg
7.6kg
7.9kg

Installation screws (Hexagon socket head 4 of M6-12)

ӖModel/Option

Stereomicroscope

60

CAD
2D

Device Unit Fixed Type: E1000

Optical fiber alignment

ӳUnit for the device holder
Can be specified the height for each 1mm.

Dimensional outline drawings

ӖE1000-h
50

16 16
40
70

2-M6 slotted hole for bolt

32
40
47.5

25

6-M3 6DP

ｈ

5.5

80
Key

E1000-h10

3.5

Guidance

SPEC

WG
alignment

Model
E1000-hHeighth15.5mm
Base plate
A24-7
Main materials٬Surface processing
Aluminium‫٭‬Black alumite treatment
Accessory: Installation screws (Hexagon socket head 2 of M6-12)

LD/PD
alignment
Alignment
components
Manualalignment unit

CAD
2D

Fiber Holder: F260
ӳThis holder is to fix the core wire.
ӳAvailable for 250μm and 900μm of jacket's radius.

Motorizedalignment unit

Dimensional outline drawings

Device unit

ӖF260-L
Fiber jacket part V groove for holding

Fiber clad part V groove for holing40

32

Fiber holders
Device holders

Sheath part V groove for holing
11

Adaptors
Contact
sensing meter

34

ӖMoldel/Option

L
F260 - 1
2 R

55

3

6

Clad diameter
Jacket diameter
Fixing type
WeightMain materials٬Surface processing

SPEC
F260-1
F260-2
L
R
L
R
125֐250μm
250μm
900μm
Magnetic force
0.1kg
Aluminium‫٭‬White alumite treatment

33

21

Model

2-M3 slotted hole for bolt

F260-1L

8

66

Stereomicroscope
Lens tube
CAD
2D

Fiber Holder for Proximity: F263

CCD
cameraӖF263-L

Monitor

When the ﬁber is mounted

12

10
50μm

125μm

21 12 2151

Detailed holding partF263-2L

5°16

ӖModel/Option

50

R

Clad diameter
125μm
Jacket diameter
250μm
900μm
Fixing type
Vacuuming and magnetic force
Weight0.2kg
Main materials٬Surface processing Aluminium‫٭‬White alumite treatment Stainless˥˥Fiber clad part
Accessory: 2m of silicon hose and installation screws (Hexagon socket head 2 of M3-10)

10

Pump

21

Probe
Alignment
systemFiber jacket holding part
Shift clamp
42

Model

SPEC
F263-1L
R°

Fiber clad part
Vacuum for holding

φ14

L

60

30

Fiber jacket
V groove for holding
φ0.125Fiber clad40

UV equipment
45 °

4

6

14

Vacuum pump setEV10AP.4-059

7.5

10 16

32

30

8

ӳAvailable for 250μm and 900μm of jacket's radius.
ӳRequire the vacuum pump for using.

Lighting

α3.5(Hose plug)

2-M3 slotted hole for bolt

2.5

R

Monitoring
unit

Dimensional outline drawings

42.5

ӳIt is a holder to use a core wire fiber in parallel.
The optical fiber clad is fixed with vacuum, and jacket is fixed with magnet.
In case of using two optical fibers in parallel, fiber holder for proximity of opposite hand should be placed face to face.
The optical fiber center-to center can be close minimum 125μm.
(When the clad radius is 125μm)
Space of one side and the bottom of optical fiber are available for any other elements.

F263-1L

Accessories
for alignment

Choice ofthe clamp arm direction
Selectionof a ﬁber jacket diameter
1：For φ250μm
2：For φ900μm
Type of holder

Accessory: Installation screws (Hexagon socket head 2 of M3-10)

F263-1R

2.5

F260-1R

Alignment
controller

6 10

WG
alignment

L
F263 - 1
2 R
Selection of a clamp arm direction
Selectionof a ﬁber jacket diameter
1：For φ250μm
2：For φ900μm
Type of holder

LD/PDalignment

4024

Optical Fiber Alignment
CAD
2D

Rotation Fiber Holder:F264●The positional alignment amount of the tip of optical fibers at the rotating is so small because thisholder will rotate based on a cladding of optical fiber.

Dimensional outline drawings■F264-L●Cladding is fixed by vacuum pump, and the jacket is fixed at the two points by the magnet.
●Correspond to a jacket diameter 250μm, 400μm and 900μm.

See WEB for R type external dimension

2-M3 slotted hole for bolt

Fixed optical fiber cladding vacuum.
(When the jacket rotate,also the cladding rotates.)

Guidance
340

11.5

Fiber rotation knob
Rotary clamp

8

ES6201 series.
Vacuum pump set：EV10A：P.4-059

6

14

40

●The end of the fiber will not be positioned to the rotation center when it is mounted on ES5201 and18

Rotating scale(0 degree~330 degree,
at intervals of 2 degree)32●A vacuum pump may be required when using this holder.

45 °
5
3.

Optical fiber alignment●This holder is for polarization-preserving fiber rotation.

2.5

Fiber jacket
V groove for holding
φ3.5（Hose plug）

16
39.5

0

20
60

F264-1R
Optical fiber jacket part
The vacuum may cause it
to come loose from thejacket characteristic.

Alignment
components
Manualalignment unit

3

F264-1L

φ37

80

LD/PD
alignment

36.5
12 20.5
7

12.5

33.5

22.5
50

20

21

WG
alignment

Fiber clad part
Vacuum for holding

14■Model/Option

1

Motorizedalignment unit

L
F264- 2 R
3SPEC

Device unit

Model
Clad diameter
Jacket diameter
Fixing type
Travel distance
WeightMain materials・Surface processing

Fiber holders
Device holders
Adaptors

F264-1
L

F264-2
R

L

F264-3R

L

Choice of the clamp arm direction
Selection of a fiber jacket diameter
1：For φ250μm
2：For φ900μm
3：For φ400μm
Type of holder

R

125μm
900μm
400μm
Vacuuming and magnetic force
360°(2°scale)
0.2kg
AluminiumーWhite alumite treatment Stainless※
250μm（※Fiber clad part）
Accessory: 2m of silicon hose and installation screws (Hexagon socket head 2 of M3-10)

Contact
sensing meter
Accessories
for alignment
Stereomicroscope

CAD
2D

Rotation Fiber Holder (without vacuum):F264N

Lens tube●Without vacuum on the tip.

Dimensional outline drawings

Monitoring
unit■Tilt stage mounted example

CCD
camera■F264N-L

See WEB for R type external dimension

2-M3 slotted hole for bolt
Rotating scale(0 degree~330 degree,
at intervals of 2 degree)

7 6.5

6

F264N-1L

45°

Fiber rotation knob

Lighting
F264N-1R

Probe
Alignment
systemAlignment
controller■Model/Option

WG
alignment
LD/PDalignment

4

025

34

20

1
L
F264N- 2 R
3

39.5
22.5
12 6.5
13

Pump

Fiber jacket holding part(V groove for holding)

21

F264N-2
F264N-3
L
R
L
R
Jacket diameter
900μm
400μm
Fixing type
Magnetic force
Travel distance
360°(2°scale)
Weight0.1kg
Main materials・Surface processing
Aluminum-White alumite treatment
Accessory: Installation screws (Hexagon socket head 2 of M3-10)

Model

21

φ37

SPEC
F264N-1
L
R
250μm

31.5
56.5

Choice of the clamp arm direction
Selection of a fiber jacket diameter
1：For φ250μm
2：For φ900μm
3：For φ400μm
Type of holder

2.5

3.5

Rotary clamp

UV equipment

8

32
40

Monitor

5

CAD
2D

Motorized Rotation Fiber Holder:FS266
Dimensional outline drawings

See WEB for R type external dimension

ӖFS266-L

3.5

Slitting for optical ﬁber set

45

24.5
300

0
33

0

100

21

28
φ20
13

Alignment
components

22.5 14
69

31

Manualalignment unit

1

SPEC
FS266-1
LR

FS266-2
FS266-3
L
R
L
R
Clad diameter
125μm
Jacket diameter
250μm
900μm
400μm
Fixing type
Vacuuming and magnetic force
PK523HPB-C15մ5 phase stepping motorOriental Motor Co.,Ltd.
Stepping motor
Travel distance
260°With CW and CCW limit sensor
Minimum resolution
0.247°/PulseFull
Weight0.3kg
Main materials٬Surface processing Aluminium‫٭‬White alumite treatmentմStainless˥

WG
alignment
LD/PDalignmentӖModel/Option

Model

Guidance

3

270
240

FS266-1R

15
41
Fiber clad part
Vacuum for holding

Fiber jacket
φ3.5（Hose plug）
V groove for holding 63.5
36.5
20.5
12
Stepping motor
7
Manual rotary knob

Stepping motor
Optical ﬁber clad part Optical ﬁber jacket part

8

58.5
28

43.5

ӳA vacuum pump may be required when using this holder.
ӳNot applicable ES5201 and ES6201 series.
Vacuum pump setEV10AP.4-059յController,DriverP.1-189

11.5

by the magnet.

20 4
32
40

Connector HR10A-10P-12S (HRS)
Connector HR10A-10J-12P (HRS)
Rotating scale(0degree~330degree, at intervals of 2degree)
6
2-M3 slotted hole for bolt
2.5
18 14

ӳStepping motor is mounted, and resolution is 0.247n.
ӳCorrespond to a jacket diameter 250μm, 400μm and 900μm.

FS266-1L

Optical fiber alignment

ӳThis holder is rotated polarization-preserving fiber automatically.
ӳThe positional alignment amount of the tip of optical fibers at the rotating isso small because this holder will rotate based on a cladding of optical fiber.
ӳCladding is fixed by vacuum pump, and the jacket is fixed at the two points

Motorizedalignment unit

L
FS266- 2 R
3Device unit

Select a clamp arm direction
Select a ﬁber jacket diameter
1：For φ250μm
2：For φ900μm
3：For φ400μm
Type of holder

Fiber holders
Device holders
Adaptors

˥Fiber clad part
Accessory: Motor cableD214-2-2Eմ
2m of silicon hose and installation screws (Hexagon socket head 2 of M3-10)

Contact
sensing meter
Accessories
for alignment
CAD
2D

Motorized Rotation Fiber Holder (without vacuum):FS266N
ӳWithout vacuum on the tip.

Stereomicroscope

Dimensional outline drawings

ӳNot applicable ES5201 and ES6201 series.

Lens tube
See WEB for R type external dimension

ӖFS266N-L

Connector HR10A-10J-12P (HRS)
Rotating scale (0degree~330degree, at intervals of 2degree)
6
6.5
6.5 2-M3 slotted hole for bolt 2.5

Monitoring
unit
CCD
camera

３００

8
32
40

28

4

58.5

３０

SPEC
FS266N-1
LR
250μm

FS266N-2
FS266N-3
L
R
L
R
Jacket diameter
900μm
400μm
Fixing type
Magnetic force
Stepping motor
PK523HPB-C15մ5 phase stepping motorOriental Motor Co.,Ltd.
Travel distance
260°With CW and CCW limit sensor
Minimum resolution
0.247°/PulseFull
Weight0.2kg
Main materials٬Surface processing
Aluminium‫٭‬White alumite treatment
Accessory: Motor cableD214-2-2E
Installation screws (Hexagon socket head 2 of M3-10)

12.5

Stepping motor
Manual rotary knob

Lighting
31

24
98.5

26

5

76

22.5
12 6.541

UV equipment
Pump

Fiber jacket

Probe

21

24.5

28
φ20

FS266N-1R

Model

０

43.5

３３０

FS266N-1L

Monitor

Alignment
systemAlignment
controller

ӖModel/Option

WG
alignment

1

L
FS266N- 2 R

LD/PDalignment

3

Select a clamp arm direction
Select a ﬁber jacket diameter

1：For φ250μm
2：For φ900μm
3：For φ400μm
Type of holder

4026

Optical Fiber Alignment
CAD
2D

Proximity Rotation Fiber Holder: F265

fixed with a magnet.
● Spaces that are in another direction and the bottom of optical fiber are available■F265-L

See WEB for R type external dimension

Rotating scale 2-M3 slotted hole for bolt
(0degree~330degree,
at intervals of 2degree)

30

10 16

45°
3.5

Rotation

WG
alignment

7.5
42.5
45°

Guidance

25°

Vacuum pump set：EV10A：P.4-059

Fiber jacket(V groove for holding)Fiber jacket(Stopper for runout)

LD/PDalignment

125μm

Motorizedalignment unit

F265-1R

F265-1L

φ0.125Fiber clad°
21

Fiber clad
Vacuum for holding

L

Fiber rotation knob

Detailed holding part

51

51

50μm

R

60
30

Fiber jacket holding part Shift clamp
φ14
Rotary clamp

Optical fiber clad part
Vaccum fixing

Manualalignment unit

6
2.5

12 21

22.5

φ3
7

33.5

φ3.5
（Hose plug）

14●A vacuum pump may be required when using this holder.
●Not applicable ES5201 and ES6201 series.

Alignment
components

When the fiber is mounted

32

proximate between other device and jig.
●Correspond to a jacket diameter 250μm, 400μm and 900μm.

840● Cladding is fixed by vacuum pump, and the jacket that has a rotation mechanism will

Dimensional outline drawings

10● The positional alignment amount of the tip of optical fibers at the rotating is so small
because when the part that hold a jacket is rotating, optical fiber.

4

Optical fiber alignment●This holder can be rotated and parallel polarization-preserving fiber.

（50）

50

6

10

■Model/Option

1

L
F265- 2 R

Device unit

3

Fiber holders

SPEC
F265-1

Model

R

Select a clamp arm direction
Select a fiber jacket diameter
1：For φ250μm
2：For φ900μm
3：For φ400μm
Type of holder

F265-3

R
L
R
125μm
250μm
900μm
400μm
Vaccuming and magnetic force
360°(2°scale)
0.2kg
Aluminum-White alumite treatment Stainless※（※Fiber clad part）
Accessory: 2m of silicon hose and installation screws (Hexagon socket head 2 ofM3-10)

Device holders

L

F265-2
L

Clad diameter
Jacket diameter
Fixing type
Travel distance
WeightMain materials・Surface processing

Adaptors
Contact
sensing meter
Accessories
for alignment
Stereomicroscope

CAD
2D

Fiber Array Holder: F267
●Holder for fixing the fiber array.
●Fix the fiber array with mechanical clamp from upper.
●The only 3~8mm fiber array can be fixed. Other sizes may be mounted with any parts

Monitoring
unitchange.

CCD
camera■Example of mounting
the fiber array

Dimensional outline drawings■F267-L

See WEB for R type external dimension

Fiber jacket
Part of holding
32
40

Lens tube

MonitorLighting

2-M3 slotted hole for bolt

30

Array holding plate18

Device clamp partUV equipment

1〜3(Applied device thickness)

Clamp knob

L

Fixing type
WeightMain materials・Surface processing

Alignment
controller
WG
alignment■Model/Option

LD/PDalignment

F267 - RR
Mechanical clamp
0.09kgAluminum-White alumite treatment
Accessory: Installation screws (Hexagon socket head 2 of M3-10)■Compatible with fiber array sizeLt
Select a clamp arm direction

4027

Type of holder

L ： min3mm
W ： 3〜8mm
t ： 1〜3mm

W
L

50
58

8

23
8

2.5

F267

Model

Alignment
system

6

SPEC

Probe

3〜8(Applied devicet width)

12.5

Pump

F267-R

21

F267-LCAD
2D

FC Connector Holder: F261
●This is a combination of adaptor F16 for a connector and bracket A54 for element adaptor.
●There are two types, one with sheath hold part and the other without sheath hold the part.
●The ferrule end-face is positioned off approx.5mm

Optical fiber alignment●Holder for fixing the fiber with FC connector.

Dimensional outline drawings■F261B-1

See WEB for F261A-1 external dimension
2-M3 slotted hole for bolt

the receptacle when a FC connector is inserted.
F16：P.3-018、A54：P.4-038

32■Connector mounting example

30

20.5

F16-2

Guidance

7.5

WGalignment

21

27.5

SPEC
F261A-1
F261B-1
Fiber with FC connector
F16-2
F16-2
No
Yes
0.06kg0.07kgAluminiumーAlumite finishing
Accessory: Installation screws (Hexagon socket head 2 of M3-10)

Alignment
components

5

2.5

Model
Applicable fiber
Adoptor
The hold part for sheath
WeightMain materials・Surface processing

LD/PDalignment

41

F261A-1

6

F261B-1

1.5

8

54.5

8

Manualalignment unit

40

62.5

Motorizedalignment unit

■Model/Option

Device unitA

F261 B - 1

Fiber holders
Select the hold part for sheath
A：N/A

Device holders

B：A

Type of holder

Adaptors
Contact
sensing meter

CAD
2D

Ferrule Holder: F262
●Holder for fixing the optical fiber with ferrule.
●This is a combination of adaptor F15 for ferrule and bracket A54 for element adaptor.

Dimensional outline drawings●There are two types, one with sheath hold part and the other without sheath hold the part.
●Non-standard hole diameter is available. Please contact us for more information. Ferrule into the hole

■F262B

Accessories
for alignment
Stereomicroscope

See WEB for F262A external dimension
2-M3 slotted hole for bolt

and tentatively fasten it using an Allen wrench. It is fixed by surface contact without damaging ferrule.
F15：P.3-018、A54：P.4-038■Connector mounting example

Lens tube

Ferule fixation screw(M3)32

Monitoring
unit

20

φd

CCD
camera

φ1.8

7.5

30

36
21

Lighting

F262A-1.8

SPEC
Model
F262A-1.8 F262A-2.5 F262A-3.0 F262B-1.8 F262B-2.5 F262B-3.0
Applicable diameter（φd） 1.8mm
2.5mm
3.0mm
1.8mm
2.5mm
3.0mm
AdoptorF15-1.8
F15-2.5
F15-3.0
F15-1.8
F15-2.5
F15-3.0
The hold part for sheathNo
Yes
Weight0.06kg0.07kgAluminiumーAlumite finishing
Main materials・Surface processing
Accessory: Installation screws (Hexagon socket head 2 of M3-10)

8

54
62

8
40

8

2.5

F262B-1.8

6

2.5

6

21

27.5

41

Monitor40

UV equipment
Pump■MountingFerrule into the hole and tentatively fasten it
using an Allen wrench.
It is fixed by surface contact without damaging
ferrule.

Allen wrench

Probe
Ferrule adaptor

Alignment
systemAlignment
controller■Model/Option

1.8
A - 2.5
F262 B
3.0

WG
alignment

Optical fiber with ferrule

Select the ferrule diameter (mm)1.8、2.5、3.0
Select the hold part for sheath
A：N/A、B：A
Type of holder

LD/PDalignment

4028

Optical Fiber Alignment

Fiber Holder with Lens: F268
Optical ﬁber alignment

CAD
2D●Holder for fixing the cylindrical elements such as optical fiber with a lens.
●This is a combination of adaptor F510C for a lens and bracket A54 for element adaptor.
●Please choose a compatible with element diameter (φd) when you order.
F510C：P.3-016、A54：P.4-038■F268-dGuidance

F268

2-M3 slotted hole for bolt
M3 fixation screw

WG
alignment

SPEC
Applicable diameter（φd）
Adoptor
WeightMain materials・Surface processing

Alignment
components
Manualalignment unit

F268-dφ4〜φ14.5mm（every 0.1mm）
F510C-d0.08ｋｇAluminiumーAlumite finishing φd part: No treatment

32

Model

LD/PD
alignment

30

Accessory: Installation screw（hexagon socket head cap bolt 2 of M3-10）

20

7.5

+0.3
φd +0.2

41■Model/ Option
27.5

F268- dFiber holders

Applicable diameter d

Device holders

Type of holder

every 0.1mm

6

4〜14.5mm

21

Device unit

6

54
62

8

8

2.5

Motorizedalignment unit

40

Adaptors
Contact
sensing meter

Base Plate for Preset Holder: E300

Accessories
for alignment■Example

Stereomicroscope

CAD
2D

Lens tube●The preset holder installed on the base plate.
●Install to each stage unit.

Monitoring
unit■Mounting■E300-L

MonitorE300-R

Pump
Probe

3
Positioning pin

SPECE300-L
E300-R
E310/ E330/ E340
Mechanical clamp
0.05ｋｇAluminiumーAlumite finishing

Clamp

Accessory: Installation screw（hexagon socket head cap bolt 2 of M3-10）

Alignment
system

V groove holder

Alignment
controller
WG
alignment■Model/ Option

LD/PDalignment

L
E300 - R
4

029

Clamp

Select a clamp arm direction
Type of holder

6
2.5

2-M3 slotted hole for bolt

25

Model
Applicable holders
Mounting
WeightMain materials・Surface processing

6.5

40
32

UV equipment

40
20

Positioning pin

12.5
25

16.5

E300-L

Lighting

See WEB for R type external dimension

8

CCD
cameraφ6

11

Preset Fiber Holder: E310
Optical fiber alignment

CAD
2D●Preset fiber holder.
●Install to the base plate E300.

■E310-1L

Fiber clad
V groove for holding

23

E310-1R
19
23

Fiber jacket
V groove for holding

Guidance
WG
alignment

25

11

10.5

13.5

5.5

5

5

13.5

10.5

25

Model
Clad diameter
jacket diameter
Mounting
WeightMain materials・Surface processing■Model/OptionL
E310- 21 R
Select a clamp arm direction

SPEC
E310-1L/R

LD/PD
alignment

13.5

E310-1L

Fiber clad
V groove for holding

19

Fiber jacket
V groove for holding■E310-1R

E310-2L/R

Alignment
components
Manualalignment unit
Motorizedalignment unit

125μm
250μm

900μm
Magnetic force
0.01ｋｇAluminum-White alumite treatment

Device unit
Fiber holders

No surface treatment of V groove part

Select a jacket diameter
1：250μm、2：900μm

Device holders

Holder model

Adaptors
Contact
sensing meter

Preset V Groove Holder: E330

Accessories
for alignment

CAD
2D

Stereomicroscope●Preset V groove holder for cylindrical elements.
●Install to the base plate E300.
●Please choose a compatible with element diameter (φd) when you order.

■E330-d-L
3.5

3.5

Monitor

4.8（MAX 13）

4.8（MAX 13）

SPEC

L

Select a clamp arm direction
Applicable diameter d
1〜5mm every 0.1mm
Holder model

UV equipment

Model
Applicable diameter（φd）
Mounting
WeightMain materials・Surface processing

Pump

Knob for clamping

Knob for clamping

E330 - d - R

Lighting

22.5

19

19

22.5

5.5

25

25

■Model/Option

16

CCD
camera3.5

16

E330-d-R

Monitoringunit■E330-d-R
3.5

5.5

E330-d-L

Lens tube

E330-d-L/R
φ1〜5mm（every 0.1mm）
Mechanical clamp
0.01ｋｇAluminiumーAlumite finishing

Probe
Alignment
systemAlignment
controller
WG
alignment
LD/PDalignment

No surface treatment of V groove part

4030

Optical Fiber Alignment
CAD
2D

Preset Fiber Array Holder: E340
Optical fiber alignment●Install to the base plate E300.
●Please choose a fiber array size when you order.
■Mounting■Example

Guidance
WG
alignment
LD/PDalignment
Alignment
components
Manualalignment unit

E340L

Motorizedalignment unit

Dimensional outline drawings

E340R■E340L-1-A■E340R-1-A
16

3.5
3.5

3.5

19
2

16

3.5

5.5

Fiber holders

19

Device unit

5.5

22.5

Adaptors

2

22.5

Device holders

Contact
sensing meter
25

Accessories
for alignment

25
Knob for clamping

Knob for clamping■Fiber array size

5.5（MAX 11）

SPEC

5.5

Monitoringunit

3.5〜4.5

Lens tube

3.5〜4.5（MAX 11）

Stereomicroscope

CCD
cameraMonitorLighting
UV equipment
Pump
Model
F.A. size
Mounting
WeightMain materials・Surface processing

Probe
Alignment
systemAlignment
controller■Model/Option

WG
alignment

L-2 -B
E340 R

E340L/R（see the picture on the right）
Mechanical clamp
0.01kgAluminiumーAlumite finishing

1 A

Select a F.A（W）
A：3〜5mm B：5〜8mm

4

031

L
t
W

※Please ask us for detail of the size.

3 C

LD/PD
alignment

Please use a fiber array bellows size or less.

C：8〜12mm

Select a F.A（H）
1：1〜1.5mm 2：2〜2.5mm
Select a clamp arm direction
Holder model

3：3〜3.5mm

L : min 3mm
W : 3〜12mm
t : 1〜1.5, 2〜2.5, 3〜3.5mm

CAD
2D

Waveguide Holder: F270/F271

Optical fiber alignment●The waveguide that hard to hold is held by vacuum.
As the hole for vacuuming can be covered by a volt and resin ring, absorption
dimension to be changed to suit the waveguide range.
●There are two types, 10mm and 20mm width.
●There are two types, one with bottom plate and the other without bottom plate.
●Custom sizes are available.
●Mechanical clamp also available.
●In your actual use, need a vacuum pump.
Vacuum pump set：EV10A：P.4-059

F270-10.20●As F271 will be fixed with key for a holder(AS49-5), it can be slid in the key direction.
●Can be used it for alignment unit bycombining with an exclusive adapter（F270PB10/20) to change optical axis height.

Guidance
WG
alignment

F271-10.20

Holder adaptor P.4-038

Dimensional outline drawings■F270-10
11

12

6-M2 cap bolt (Vacuum hole)11122-φ1.6 (Vacuum hole)

Mechanical clamp

11

11
Mechanical clamp

1.5 5 1111

φ3.5 (Hose plug)

3-M2 3DP

3-M2 3DP

11

1

1
2
14

30

Device holders
Adaptors

6

30
6
14
40

Fiber holders

2

1.5 5 11

Slide lock screw (M4)

Device unit

20

φ3.5 (Hose plug)

Slide lock screw (M4)14.5

14 14.5
40

14

Contact
sensing meter■F271-20
11

12

11

4-3.5THRUφ6 C'BORE 3.5DP(M3 bolt hole)

6-M2 cap bolt (Vacuum hole)
2-φ1.6 (Vacuum hole)1112

3-M2 3DP

φ3.5 (Hose plug)

11
32
40

Mechanical clamp

Stereomicroscope
Lens tube

3-M2 3DP
1.5
5
14

11

11
32
40
1

2

11

1

1.5
5
14

Accessories
for alignment

6 6
20

40
32

10

40
32

Mechanical clamp

φ3.5 (Hose plug)

4-3.5THRUφ6 C'BORE 3.5DP(M3 bolt hole)

112■F271-10
3-M2 cap bolt (Vacuum hole)
φ1.6 (Vacuum hole)

Motorizedalignment unit

6 6

10

φ1.6 (Vacuum hole)

Alignment
components
Manualalignment unit

■F270-20

3-M2 cap bolt (Vacuum hole)

LD/PDalignment

Monitoring
unit
CCD
cameraLighting

6

6

36

36

MonitorSlide lock screw (M4)

Slide lock screw (M4)■How to mount a vacuum adsorption.

Waveguide
Waveguide holder

Remove a provided screw
with hex wrenchwhen vacuuming
hole is opened.

UV equipment
Attached screws
Ring

Pump

4.5

Probe
Alignment
system

To the vacuum pump

Alignment
controller
WG
alignment

SPEC
Model
F270-10F270-20
F271-10F271-20

Absorption area
10×40mm
20×40mm
10×40mm
20×40mm

Main materials・Surface processing
Weight0.02kgVacuuming +Aluminum0.04kg
Mechanical clamp
Black alumite
0.06kg（Side）
treatment
F271-B
0.08kgAccessory: 2m of silicon hose and installation screws (Hexagon socket head 4 of M3-6)※Included screws for only F271-10 and F271-20
hold operation

Bottom plate

LD/PDalignment

4032

Optical Fiber Alignment
CAD
2D

Waveguide Holder: F272/F273
Optical fiber alignment●Hold of waveguide less than 10mm in length. Consider protruding fiber from both sides.
●There are two types, one with bottom plate and the other without bottom plate.
●The clamp is a vacuum method and machine clamp combination with the springs.
Vacuum pump set：EV10A：P.4-059
Can be used it for alignment unit by combining
with an exclusive adapter（F270-PB10) to change optical axis height.

Guidance
WG
alignment

F272

LD/PD
alignment

Dimensional outline drawings

Alignment
components■F272-2

F273

Holder adaptor P.4-038■F272-5

14.5
Clamp knob
63
62

6

11.5 1440

Slide lock screw (M4)

Adaptors

5

Vacuum section

6

20
30

31

Vacuum section

5

20
30

2

2

Device clamp part
Vacuum hole (φ2)

10 12

5

Fiber holders
Device holders

64.5
62

Device clamp part
Vacuum hole (φ1)

10 12Device unit

20.5
Clamp knob

φ3.5 (Hose plug)31

φ3.5 (Hose plug)

Motorizedalignment unit

10

5

10

2

Manualalignment unit

10

11.5 1440

Slide lock screw (M4)

27
67

10

32
72

Contact
sensing meter■F273-2■F273-5
4-3.5 THRU φ6 C'BORE 3.5DP（M3 bolt hole）

4-3.5 THRU φ6 C'BORE 3.5DP（M3 bolt hole）

Stereomicroscope

Monitoring
unit

φ3.5 (Hose plug)

CCD
camera

2

32
10

40
63
62

14.5
Clamp knob
Device clamp part

5

40
32
10

5

40
32
10

2

Lens tube

32

φ3.5 (Hose plug)

10

64.5
62

Vacuum section

6

6

UV equipment

5

37

26
36

37

Lighting

2

2

10 12

5

Clamp knob
Device clamp part
Vacuum hole (φ2)

10 12Vacuum hole (φ1)

Monitor

20.5

40

26
36

Accessories
for alignment

Slide lock screw (M4)

Pump

Slide lock screw (M4)

Probe
Alignment
systemAlignment
controller
WG
alignment
LD/PDalignment

4

033

Model
F272-2
F272-5
F273-2
F273-5

Device width
2〜5mm
5〜10mm
2〜5mm
5〜10mmSPEC
hold operation
Bottom plate

WeightMain materials・Surface processing
0.04kgAluminum0.04kg
Vacuuming +Black alumite
Mechanical clamp
0.08kgtreatment
F271-B
0.08kgAccessory: 2m of silicon hose and installation screws (Hexagon socket head 4 of M3-6)※Included screws for only F273-2 and F273-5

CAD
2D

Preset Waveguide Holder: E400

Optical fiber alignment●Waveguide holder preset type.
●Fix waveguide by vacuum.
●As the hole for vacuuming can be covered by a volt and resin ring,absorption dimension to be changed to suit the waveguide range.
●Install to the preset base plate E410-B.

■Example

Guidance
E400-6E400-20

E400-40

WGalignment

Vacuum pump set：EV10A：P.4-059

LD/PDalignment

Dimensional outline drawings

10
40

3.5

Device unit
Fiber holders

3-M2 3DP

1.5 5 11 1142

3-M2 3DP2

φ3.5（Hose plug）

1

1

24.5

8
40
1.5 5 11 1142

Manualalignment unit
Motorizedalignment unit

12

12

φ3.5
（Hose plug）

2

3-M2 3DP

12 11 11 3-M2 cap bolt
φ1.5
（Vacuum hole） 3.5（Vacuum hole）
Mechanical clamp

2

40
20

12
φ1.5
1.5 5 11 11（Vacuum hole）
42

1

φ3.5
（Hose plug）

3.5

Mechanical clamp

24.5
8
6

3.5

3.5

12 11 11 3-M2 cap bolt
φ1.5
（Vacuum hole）（Vacuum hole） 3.5
Mechanical clamp
24.5

3-M2 cap bolt
（Vacuum hole） 12 11 11■E400-1020■E400-820■E400-6

Alignment
components

Device holders

42

8

14

10

4

70
10

10

70

70

Adaptors

42

14

Contact
sensing meter
Accessories
for alignment

42

Stereomicroscope

15 10
30
40

12 24.5
5 10 10 10
φ3.5（Hose plug）

1.5 5 11 1142

12 11 11
3.5

12-M2 cap bolt
（Vacuum hole）
Mechanical clamp

Monitoring
unit
CCD
camera20
40

3.5

4-φ1.5（Vacuum hole）

Lens tube

Monitor10
1.5 5 11 1142

6-M2 3DP2

2

6-M2 3DP

φ3.5（Hose plug）

2

3-M2 3DP

9-M2 cap bolt
（Vacuum hole）
Mechanical clamp

12 11 11
3.5

1

1.5 5 11 11
42
1

φ3.5
（Hose plug）

3-φ1.5（Vacuum hole）
3.5

20
40

6-M2 cap bolt
（Vacuum hole）
Mechanical clamp20

12 24.5
14 12

3.5

2-φ1.512 11 11
（Vacuum hole） 3.5■E400-40

1■E400-30

12 24.5
10 10 10■E400-20

Lighting
UV equipment

Model
E400-6E400-8E400-10E400-20
E400-30
E400-4014

42

Absorption area
6×38mm
8×38mm
10×38mm
20×38mm
30×38mm
40×38mm

70
10

70
10

70
10
14

SPEC
hold operation
Vacuuming +Mechanical clamp（Side）

Pump

14

42

Probe

42

Alignment
systemAlignment
controller

Weight0.09kg0.10kg0.11kg0.18kg0.25kg0.32kg

WGalignment

Main materials・Surface processing

LD/PDalignment

AluminumBlack alumite treatment

4
Accessory: 2m of silicon hose034

Optical Fiber Alignment
CAD
2D

Preset Waveguide Holder (Mechanical clamp type): E400
Dimensional outline drawings■E400-10N■E400-20N

3.5

38.5

3.5

38.5

3.5

38.5

15

42
57

15

42
57

15

42
57

E400-N

Alignment
components

Model
E400-8N
E400-10N
E400-20N

Manualalignment unit
Motorizedalignment unit

Device attached surface
8×38mm
10×38mm
20×38mm20
40
70■Details of the mechanical clamp
Available replacement
5mm to the rightSPEC
Hold operation
Mechanical clamp

Weight0.10kg0.12kg0.18kg2

1.8（〜3.8）

LD/PDalignment

10

10

10

WG
alignment

2

2
70

2
70

Guidance

12 24.5

12 24.5840

12 24.53.5

3.5■E400-8N
3.5

Improved holding power by mechanical clamp.
●No vacuum function.
●Install to the preset base plate E410-B.

1040

Optical fiber alignment● Add the clamp from the upper side to E400.

5Main materials・Surface processing
AluminumBlack alumite
treatment
5

Device unit
Adjustable 2mm of mechanical clamp
height adjustment screw

Fiber holders
Device holders

CAD
2D

Preset Waveguide Holder Unit: E410

Adaptors●This holder is combined with E410-B and E400.
●When a variety of element size, should be changed only E400 to match the size
what you want.
●Install to the stage unit E1000.

Contact
sensing meter
Accessories
for alignment
Stereomicroscope

Dimensional outline drawings■E410-6■Example for using

Lens tube

Waveguide
WG holder

Monitoring
unit

E400

CCD
cameraLighting

E410-6

Preset baseE410-B※Not to be attachable
to the tilt stage
Device unit E1000(No key）

80

MonitorBase
20

UV equipment
Pump

40

Probe
Model
E410-6
E410-8E410-10E410-20
E410-30
E410-40

Alignment
systemAlignment
controller
WG
alignment
LD/PDalignment

4

035

Model
E410-8N
E410-10N
E410-20N

SPEC
Main materials・Surface processing
Absorption area
hold operation
Weight
6×38mm0.14kg
8×38mm0.15kgVacuuming +10×38mm0.16kgAluminiumー
Mechanical clamp
Alumite finishing
20×38mm0.23kg（Side）
30×38mm0.30kg40×38mm0.37kgAccessory: 2m of silicon hose and installation screws (Hexagon socket head 4 of M3-10)
Device attached surface
8×38mm
10×38mm
20×38mmHold operation

WeightMain materials・Surface processing
0.15kgAluminiumー
Mechanical clamp
0.17kgAlumite finishing
0.23kgAccessory: Installation screws (Hexagon socket head 4 of M3-10)

9

48.5

4

CAD
2D

Temperature Control Waveguide Holder: F274
Dimensional outline drawings

ӳA mechanical clamp also available.
ӳThe holder will be changed to the thermal displacement when temperature

ӖF274-18
2

3-3.5 THRU
（M3 bolt hole）

control width is getting wider. Please ask for more details.

16

16

8

Mechanical clamp

φ3.5（Hose plug）

6.5

11
12

11

3-M2 3DP

11

Guidance

1

11

9
3.5

11
18
22
26

ӳBuilt in the corrected 10kΩ thermistor sensor using the Peltier device.
Vacuum pump setEV10AP.4-059

Optical fiber alignment

ӳThe waveguide holder with temperature controller.
ӳ As the hole for vacuuming can be covered by a volt and resin ring,absorption dimension to be changed to suit the waveguide range.

36

WGalignment
15

LD/PD
alignment
Alignment
components

47
7.5

67

Manualalignment unit

Connector
9-pin D-sub
1
2
3
4
5
6
7
8
9

F274-18SPEC
Model
F274-18
Absorption area
18×37mmHolding
Vacuuming(Tubing size IDĮ3.0mm ODĮ5.0mm) + Mechanical clamp(side)
Peltier devices
13Wյ4Vյ5A
Thermistor sensor
VTS510 (Calibrated 10kΩ thermistor sensor)
Weight0.1kg
Aluminum-Black alumite treatment
Main materials٬Surface processing
Accessory: 2m of silicon hose and installation screws (Hexagon socket head 3 of M3-18)

Peltier devices

Motorizedalignment unit

Peltier devices

Device unit
Fiber holders

Thermistor sensor
Thermistor sensor

Device holders
Adaptors
Contact
sensing meter
CAD
2D

Objective Lens Holder: F280/F281

Monitoring
unit

20
M20.32 P=0.706

CCD
cameraMonitor

14.5

Lighting
M4 Slide lock screw

15
40

10

ӖF281

UV equipment
40
32

Pump

4-3.5THRUφ6 C'BORE 3.5DP（M3 bolt hole）

Probe

SPEC
Model
F280
F281
Bottom plate−
F271-B
Weight0.02kg0.06kgAluminum-Black alumite treatment
Main materials٬Surface processing
Accessory: Installation screws (Hexagon socket head 4 of M3-6)
˥Included screws for only F281

32

40

12.5

F281

20
M20.32 P=0.706

Alignment
systemAlignment
controller

10

36
24.5

51

WG
alignment
LD/PDalignment

11.5

9

M4 Slide lock screw

5

6

Holder adaptor
P.4-038

ӖF280

30

ӖLens mounting example

Lens tube

3

F280

Dimensional outline drawings

45

Can be used it for alignment
unit by combining with an
exclusive adapter(F270PB10) to change optical axis
height.

Stereomicroscope

6

ӳShould be used JIS standard screw M20.32P=0.706 for fixing objective lens.
ӳThere are two types, one with bottom plate and the other without bottom plate.

Accessories
for alignment

4036

Optical Fiber Alignment
CAD
2D

Selfoc® Lens Holder: F290
Optical fiber alignment●This holder is combined with Selfoc lens holder F14 and bracket A54-0.
F14：P.3-015、A54：P.4-038

■F290

30

32

B

14

2-M6 slotted hole for bolt

Guidance

M4 fixation screw
φ10

φA

8
40

2.5

Alignment
components

3F290-1

Manualalignment unit

φ20
φ10

67
70

20

6
10

21

LD/PD
alignment

21

41

WG
alignment

4.5
11
18

48
66

90

SPEC

Motorizedalignment unit

Model
F290-1
F290-2
F290-3F290-4F290-5F290-6

Device unit
Fiber holders

Applicable lens diameter (A)φ0.5〜 1.5mm
φ1.6〜 2.5mm
φ1.6〜 2.5mm
φ2.5〜 5.0mm
φ4.0〜 8.0mm
φ7.0〜12.0mm

Width (B)hold operation

WeightMain materials・Surface processing

Mechanical clamp

0.07kgAluminiumー
Alumite finishing

2mm

4mm

Accessory: Installation screws (Hexagon socket head 2 of M3-10)

Device holders
Adaptors
Contact
sensing meter
Accessories
for alignment
Stereomicroscope

LD Holder (TO-Package type): F125

Lens tube●Built-in Peltier device holder.

Monitoring
unit
CCD
cameraMonitorLighting
UV equipment
Pump
Probe
Alignment
systemF125

Alignment
controller
WG
alignment
LD/PDalignment

4

037

SPEC
Model
F125-3A
F125-3B
F125-4A
F125-4B
LD Size
φ5.6
φ9.0
φ5.6
φ9.0
Pin P.C.D.
2.0mm
2.54mm
2.0mm
2.54mmNumber of pin
3
4
Weight0.2kg
Main materials・Surface processing Aluminum-Black alumite treatment Cooper-Gold plating processing
Detailed LD holder P.3-035

CAD
2D

CAD
2D

Braket for Element Adaptor: A54

Optical fiber alignment●Available for variety of adaptors.
●Tilt stage-mountable.
●A54-0 type, no jacket hold part, and A54-1, has it.

Available in various adaptors (except F19) as below the picture.
(Use M2.6-P.C.D.32 screw for mounting adaptor)
P.3-015〜018

Guidance
A54-0

A54-1

WG
alignment

Dimensional outline drawings■A54-0LD/PDalignment■A54-141.5

32Manualalignment unit

142-M3 6DP
16

21
2.5

10

6
2.5

8
32
40

Fiber holders
Device holders

2.5

48
56

Device unit

R1

2.5

R1

8
32
40

8

15

16

21

15

6

10

17.5

2-M3 6DP

48

Motorizedalignment unit

D.3
2

30

7.5

P.C
.

2-M2.6 6DP

14

2-M2.6 6DP

P.C
.

D.3
2

32
30

Alignment
components

2 bolt holes for 2-M3

2-M3 slotted hole for bolt

Adaptors
ModelPart of holding
WeightMain materials・Surface processing

SPEC
A54-0
A54-1
No
Yes
0.04kg0.06kgAluminum-White alumite treatment
Accessory: Installation screws (Hexagon socket head 2 of M3-10)

Contact
sensing meter
Accessories
for alignment
Stereomicroscope
CAD
2D

Holder Adaptor

Lens tube
Monitoring
unit●This is for the device holder. It is designed to be 80mm optical axis height by the mounting device holder (F270~F274, F280, F281) and adaptor (F270-PB10/20/26).
●In order to avoid the interference of the alignment unit.
●Also has line-up the base plate (F271-B, E410-B).
F270-PB10

F270-PB20

CCD
cameraSee WEB for external dimension

F270-PB26

Monitor■Example for using

Lighting
UV equipment
Pump
Probe

F271-B

Model
F270-PB10
F270-PB20
F270-PB26
F271-B

Alignment
system

E410-BSPEC
Applicable holders
WeightF270-10 F270-20 F271-10 F271-20 F274-18 F272-2 F272-5 F273-2 F273-5 F280
F281
○
○
○
○
○
○
○
○
0.06kg○○
0.1 kg○
0.1 kg○○○○○0.04kgMain materials・Surface processing：AluminiumーAlumite finishing
Accessory: Installation screws Hexagon socket head 4 of M3-6
（F271-B）
4 of M3-10（E410-B）

（

）

Model
E410-B

Applicable holders
E400 series

Weight0.06kgAlignment
controller
WG
alignment
LD/PDalignment

4038

Optical Fiber Alignment
CAD
2D

2-axis Tilt Table: EB55
Optical fiber alignment●This is the 2-axis tilt control table for matching a fiber holder parallel with the optical axis.

Dimensional outline drawings

Good for proximity to the probe and monitoring.

■EB55-1L

4-M4 C'BORE 5

50
5
888 4

4
10

32
40

60
50

4
10

5

8

32

5

10-M3 4.5DP

EB55-1L

21

EB55-1R±2.5°
Vertical approx.0.57°/per rotation
Horizontal approx.0.58°/per rotation
0.28kgAluminum-Black alumite treatment60■EB55-1R

10-M3 4.5DP

EB55-1 L
R

Motorizedalignment unit

50
60

40
32

■Example for using■Model/Option

10
4Manualalignment unit

10
4

Accessory: Installation screws (Hexagon socket head 4 of M4-6)※Value of near the center of stroke

Alignment
components

5 4-M4 C'BORE
550
54 888 8

10

5

LD/PD
alignment

Center of rotation

32

WG
alignment

SPEC
EB55-1L

Model
Travel distance
Travel distance per
one rotation(※）
WeightMain materials・Surface processing

60

Holder table height

31
29

Guidance

35.4

EB55-1R60

35.4

Center of rotation

Holder table height

29
31

21

L：Left type R：Right type

Device unit
Fiber holders

1060

Device holders
CAD
2D

Tilt Stage: B50/B51/B53

Adaptors●This is the 2-axis tilt stage for mouting a fiber holder.
Selectable from 3type due to type of rotation center.

Contact
sensing meter
Accessories
for alignment●Mount to the high resolution stage B10 series.
B50-60RN

Stereomicroscope

B51-60RN

Detailed High resolution stage B10 P.2-089
Alignment unit P.4-017
See WEB for external dimension

B50-60LN

Lens tube

B51-60LN●There are 3 types of different type of rotation center.
B50: Inner tilt stage (Center of the rotation is the inside a stage)

Monitor

B51: Outer tilt stage (Center of the rotation is the outside a stage)
B53: Side tilt stage (Center of the rotation is the side a stage)

Lighting●It is very important to make a relation of tilt, rotation center and device position (e.c.
fiber edge) when you wish a fine motion to begin.
As figure A shows, there is almost no difference in the top of optical fiber when it isaligned with the tilt rotation center.
As figure B and C shows, there may be a discrepancy in location at the top of opticalfiber caused by the gap between the top of optical fiber and tilt rotation center.

UV equipment
Pump
Probe

●This is the control mechanism for positioning horizontal optical axis and optical fiber.
Do not use in vertical and upside down.

Alignment
systemSPEC
Model
B50-60LN B50-60RN B51-60LN B51-60RN B53-60LN B53-60RN
Stage surface size
40×60mm
Travel distance±3.0°（θy±3.047° θz±3.013°） ±2.5°（θy±2.645° θz±2.514°） ±2.5°（θy±2.645° θz±2.869°）
Micrometer minimum
θy33.8〃/scaleθy29.3〃/scaleθy29.3〃/scalereadable scaleθz33.4〃/scaleθz27.8〃/scaleθz31.8〃/scale
Weight0.32kg0.39kg0.40kgMain materials・Surface processing
Aluminum-Black alumite treatment

Alignment
controller
WG
alignment
LD/PDalignment

Accessory: Installation screws (Hexagon socket head 4 of M4-6)

4039

Dimensional outline drawings
Center of rotation

Center of rotation

20

Center of rotation

20

21

CCD
cameraB53-60RN

21

B53-60LN

21

Monitoring
unit

10

5

B50（Inner tilt）
Center of rotation

9

10

B51（Outer tilt）

B53（Side tilt）

Optical fiber

Fig.A

Fig.B

■Model/Option0

B5 1 -60 L
RN
3

L：Left type R：Right type
0：Inner tilt、1：Outer tilt、3：Side tilt

Fig.C

Contact Sensing Meter: CSM
Con
Optical fiber alignment

control or facing (paralleling) of fiber array and waveguide.
ӳCSM for GAPP conӳ
gnm unit E/ES series.
ӳMount for the alignmentӳ
for GAP adjustment/faing.
le foӳCSM makes it possibleӳӖExample
Guidance
WG
alignment

17
1
7

LD/PD
alignment
Alignment
components
Manualalignment unit
It might be slig
slightly different from the actual product.

ӖFeaturesӖMotorizedalignment unit

Ӳ enables positioning while checking a display.
ӲItning while checking a display.
ӲAs the contact position is detected, it anables positioningӲ
reat importace to work efficiency.
ӲӲProvides a multi-device holder which attaches great

Device unit
Fiber holdersSPEC
A measuring device main body and one set of L and R holder
3֐12W×3֐20L×1֐3.5mmtContact realizationGAP adjustmentFacing
Contacting position w/in ±2μmյFacing w/in ±0.05°
AC100֐240V±10%˥1 50/60Hz 18W
main body160W×130D×95Hmm
Holder80W×55D×20Hmm
main body1.2kg
Holder part: 0.12kg each

Configuration
Applicable device size
Measuring function
Reproducible
Electric power
Size
WeightDevice holders
Adaptors
Contact
sensing metera(22 V)˥1Power plug is different between Japan(100V) and China(220V)

Accessories
for alignment
StereomicroscopeԼքExampleօManual stages alignment unitE2000յE2200 series
Motorized stage alignment unitES3800յES series
Attachable to Manual alignment system (E8203B)
P.4-017֐

Lens tube
Monitoring
unit
CCD
cameraԷ

MonitorLighting
E8203BUV equipmentOptic PM

UV equipment

Monitor for observation

Pump
Ը

Թ

Ի

Probe

Ժ

Alignment
system

Ζ
Η
Θ
Ι
Κ
Λ

Configuration
Main bodyTouch-sensing holder L
Touch-sensing holder R
FA holder L
FA holder R
Power adaptor

Alignment
controller
WG
alignment
LD/PDalignment

4040