Barco Healthcare
Medical solutions portfolio

1About Barco
Focused on transforming the delivery of care, Barco connects healthcare professionals
at every patient touchpoint, from the imaging room to radiology, through specialist
consultations and into the surgical suite. We offer a network of medical imaging
solutions that deliver the complete picture to support more informed decisions, when
and where it matters most.
RADIOLOGY
BREAST IMAGINGDENTISTRYPATHOLOGY

SURGERY

MODALITYIMAGING
DERMATOLOGYHOME READING2015
GLOBAL MEDICAL IMAGE VISUALIZATIONVISIONARY INNOVATION LEADERSHIP AWARD

Recognized as the Visionary Innovation
Leader in medical displays by Frost &Sullivan2019
NORTH AMERICAN
MULTI-MODAL MEDICAL DISPLAY MONITORSTECHNOLOGY LEADERSHIP AWARD

Recognized as the Technology Leader inthe Multi-modal Medical Display Industry
by Frost & Sullivan

Please note that some of our products might not be available in all regions2

3

Coronis Uniti®
breast imaging

Radiology displays andbreast imaging displays

radiology

The first diagnostic display system explicitly designed for both PACS and multimodality
breast imaging. Unique in its spectacular representation of calibrated color and

12 reasons Coronis Uniti is an
excellent companion for breast
radiologists

grayscales. Remarkable in its 2D and 3D capabilities. Capable of both static anddynamic imaging.

Sometimes,
life is in the details
12 reasonsYour benefits

Coronis Uniti is an excellent companion
for breast radiologists+
Worklist display=+
Mammography displays

Color PACS displays

Coronis UnitiTM> One display for all images to unify your workflow
1

2

Cleared for viewing all typesof breast imaging exams: (3D)
mammography, breast MRI, breast
ultrasound...

Industry-first color calibration for
accurate display of color breast
images3

4> 12% more pixels to display more or larger images> The brightest display on the market - delivers exceptional brightness resulting in
more JNDs> Barco Optical GlassTM reduces reflections and improves contrast> Rapidframe technology, used in displays intended for Digital Breast Tomosynthesis,
can lead up to 10% higher detection of small details 1> Calibrated luminance levels of 1000 cd/m² lead to increased detection probability ofbreast microcalcifications compared to calibrated luminance levels of 500 cd/m² 2> Optimized for the standard field of vision and low reflection to minimize eye-fatigue

1ST

1000 cd/m²

Up to 10%Higher detection of small details
in digital breast tomosynthesis with
RapidFrame technology*
5calibrated luminance
2% increased reading accuracy using
Coronis Uniti, compared to a display
with 500 cd/m² calibrated luminance**
6

12MP

33”
2 x 5MP1 x 12 MP

18%

12%

More screen real estate forside-by-side comparisons andimage fusions

More pixels for clear viewing ofdense breast tissue dense breast
tissueRead more
7

Fusion display for reduced eye-strain
during long reading sessions***

8

9

1ST

> Intuitive workflow tools selectively dim/boost and focus light for closer inspection> Cleared for multimodality breast imaging, incl. breast tomosynthesis, 3D
mammography, breast MRI, breast ultrasound, CT and ultrasound, including
vascular and gynecological ultrasound> Industry-leading 5-year warranty

Unique intuitive workflow tools
to boost/dim illumination for
closer inspection

Accurately renders color andgrayscale for every image type - evenwhen viewed simultaneously

10

11

Barco Optical Glass for exceptional
local contrast and reduced
reflection

2x the lifetime and 2x the brightnessof other mammography displays

2x

12

2x

Reduced clicks
Up to 4 fewer mouse clicks per
screening and/or diagnostic study

Marchessoux, C., et al. (2011). Validation of a new digital breast tomosynthesis medical display. Proceedings of SPIE, 7966, 79660R
Krupinski, E. (2018). Reducing Radiation Dose in Digital Mammography by Increasing Display Luminance. Proceedings of SIIM, 2018.
***
Weschler, M. (2012). 6MP Displays Offer Speed, Comfort Benefits. Auntminnie, 2012.
***www.barco.com/healthcare

Models and options

4

Meet Coronis Uniti

5

Coronis Fusion

Niobreast imaging

radiology

radiology

dentistry

PIVOT

Coronis Fusion, perfected for color imaging with vivid and calibrated colors to help

A family of diagnostic displays that bring dependable diagnostic imaging to its true

you see important color image details. Ergonomically designed to reduce repetitive

potential. Nio’s proven technologies ensure diagnostic confidence at all times, mak-stresses. Optimized for an efficient workflow with our suite of Intuitive Workflow Tools.

ing it the standard for the radiology reading room.

Your benefitsYour benefits
Meet Coronis Fusion> Most efficient dual-head option on the market> Bright, calibrated luminance of 600 cd/m2 for the lifetime of the display> Highest brightness and contrast - luminance ratio exceeds ACR guidelines> Uniform luminance with patented Color PPU technology> Minimal image manipulations – less windowing, leveling, zooming & panning> High contrast of 2000:1 to see subtle image details> More DICOM JNDs to see more subtle details quicker> SteadyColorTM calibration technology to see details in the important color> Uniform image rendering center to edge
> Factory-fit front cover reduces reflection and improves visible contrast

content of images> Industry-leading 5-year warranty> Wide Gamut display> Wide viewing angle in your field of view> Higher brightness allows you to see more details with less eye strain> SoftGlowTM task and wall lighting> SpotView

TM

to see more details in the image more quickly – 15.5% decreased

reading time
> Auto QA and compliance tests meet the latest regional standards> Assure compliance and uptime of all your PACS workstations> Intervention-free QA saves time, reduces effort and cost
> Industry-leading 5-year warranty

Models and options

6

Watch the demo

Clinical collaboration at home, in the hospital, anywhere
Nio Color 8MP• Designed for home reading with integrated microphone, camera and speakers• Large 32” screen size with 8MP resolution and excellent image quality• Automated QA & compliance

Models and options

7

QAWeb Enterprise

Intuitive Workflow Tools

Always a clear view

Diagnostic work can be complicated, with many cases to analyze in little time. That’s
why your medical display should not just be a medium to view digital images, but also
an active tool that helps you work with them, comfortably. Our Intuitive Workflow
Tools are designed to do exactly that.
Thanks to our advanced medical display controllers, Intuitive Workflow Tools areintegrated into your Barco display for radiology, mammography, pathology, clinicalreview imaging and dentistry.
> Improve your focus with a range of tools specifically designed to help you
concentrate on important details.
> Adapt your workstation’s settings to how both you and your colleagues work best.
> Boost your workflow further thanks to those extras and make your workspaceextra comfortable!
You choose which ones you use, and how you use them. Work smarter, not harder!

Learn more about our suite of Intuitive Workflow Tools

QAWeb Enterprise supports consistent image quality through automated calibration

Some Intuitive Workflow Tools in the spotlight…

and QA, and also enables compliance to the latest regional and international regula-

SpotView

tions for image quality. Get the most out of your display fleet, in the hospital, at home,or elsewhere, and ensure best-in-class performance and optimal usage throughout
your displays’ lifetimes.

What makes QAWeb Enterprise so unique?

QAWeb Enterprise video

SpotView enables focused observation during readings by dimming images outside aregion of interest and enhancing the contrast within it.
Especially helpful when viewing pediatric extremities, breast calcifications and chest
images, SpotView supports analyzing subtle differences in the image.

DL Precise> Scalable: Powerful tools to manage workstations from anywhere, at anytime> Flexible: Customize policies to meet your quality requirements> Intuitive: Effortless compliance and smart reporting> Secure: Assured security and patient privacy

If you’re a radiologist who frequently handles dense breast images, then DL Precise
is a true game-changer. The AI-driven software tool augments an area you choosewith a vividly colored density map and calculates the morphology of suspicious areas.
DL Precise is currently available in limited regions. Please consult your Barco representative
to confirm availability.
Watch the demo

SoftGlow

Download QAWeb Enterprise8

Get full control of the ambient light surrounding your workstation. Displays equippedwith SoftGlow offer you two light sources, coming from your display.
In this way, your eye needs to adapt less intensively when it moves from your brightdisplay to something in your darker office. You can choose which one should be onor use both at the same time. Adjust their brightness according to your preference!

9

R A D I O L O G Y D I S P L AY S A N D M A M M O G R A P H Y D I S P L AY S
2MP

3MP

4MP

5MP

6MP

8MP

12MP

MDNC-2521

MDNC-3521

MDCC-4430

MDNG-6221

MDNC-6121

MDCC-6530

MDNC-8132

MDNC-12130

MDMC-12133

Nio Color 2MP

Nio Color 3MP

Coronis Fusion 4MP

Nio Gray 5.8MP

Nio Color 5MP

Coronis Fusion 6MP

Nio Color 8MP

Nio Fusion 12MP

Coronis Uniti®

5 years

5 years

5 years

5 years

5 years

5 years

5 years

5 years

5 years

433 x 325 mm (17.1 x 12.8")

433 x 325 mm (17.1 x 12.8")655 x 410 mm (25.8 x 16.1”)

324 x 433 mm (12.77” x 17”)433 x 325 mm(17 x 12.8”)654 x 409 mm(25.8 x 16.1")

708.48 x 398.52 mm(27.9 x 15.7")653 x 435 mm(25.7 x 17.1")

708.1 x 472.1 mm(27.8 x 18.6")

Screentechnology

IPS-SFT Color LCD

IPS-SFT Color LCD

IPS-TFT color LCD

LCD

a-si TFT active matrix
dual domain IPS

IPS-TFT color LCD

IPS

LCD

IPS

Diagonal size

541 mm (21.3")

541 mm (21.3")

772 mm (30.4”)541 mm (21.3”)

540 mm (21.3”)

772 mm (30.4")

812.80 mm (32.0")

784 mm (30.9")853.44 mm (33.6")

Aspect ratio

4:3

4:3

16:10

3:4

3:4

16:10

16:9

3:2

3:2

2MP (1200 x 1600)

3MP (1536 x 2048)

Native 4MP (2560 x 1600)
Configurable to 2 x 2MP+
(1280 x 1600)
Configurable to 2 x 2MP(1200 x 1600)

5.8 MP (2100 x 2800)

5.8MP (2100 x 2800)

Native 6MP (3280 x 2048)
Configurable to 2 x 3MP+
(1640 X 2048)
Configurable to 2 x 3MP(1536 X 2048)

8MP (3840 x 2160)

Native 12MP
(4200 x 2800)
Configurable to 2 x 5.8MP(2100 x 2800)

4200 x 2800

1000 cd/m² (600 cd/m²)

1050 cd/m² (600 cd/m²)

1050 cd/m² (600 cd/m²)

1000 cd/m² (600 cd/m²)

1000 cd/m² (600 cd/m²)

1050 cd/m² (600 cd/m²)

850 cd/m² (420 cd/m²)

1200 cd/m² (600 cd/m²)

2100 cd/m² (1000 cd/m²)

2000:1

2000:1

2000:1

1400:1

1400:1

2000:11350:1

1500:1

1200:1

Screen protection

Factory fit protective,
non-reflective glass cover

Factory fit protective,
non-reflective glass cover

Factory fit protective,
non-reflective glass cover

Factory fit protective,
non-reflective glass cover

Factory fit protective,
non-reflective glass cover

Factory fit protective,
non-reflective glass cover

N/A

N/A

Protective, anti-reflective
optical glass

Sensor technology

Integrated I-Guard frontofscreen photometer sensor,
ALC

Integrated I-Guard frontofscreen photometer sensor,
ALC

Integrated I-Guard front-ofscreen color sensor, Intelligent Multi-Sensor
Technology, ALC

Integrated I-Guard front-ofscreen photometer sensor,
ALC

Integrated front-of-screen
sensor, BLOS, ALC

Integrated I-Guard front-ofscreen photometer sensor,
Intelligent Multi-Sensor
Technology, ALC

Ambient light sensor,
I-Guard front sensor,
Presence sensor

I-Guard, ALS

Integrated color I-Guard
front-of-screen photometer
sensor, Intelligent MultiSensor Technology, ALC

Diagnostic imaging
technology

SteadyColor, SteadyGray,
MXRT, Intuitive Workflow
Tools

SteadyColor, SteadyGray,
MXRT, Intuitive Workflow
Tools

SteadyColor, Color PPU,
SmoothGray, MXRT,
Intuitive Workflow Tools
with optional touchpad

ULT, SteadyGray, MXRT,
Intuitive Workflow Tools

ULT, MXRT, Intuitive
Workflow Tools, SteadyColor, SteadyGray, Clear/
BlueBase

SteadyColor, Color PPU,
SmoothGray, MXRT, Intuitive Workflow Tools with
optional touchpad

ULT, MXRT, Intuitive Workflow Tools, SteadyColor,
SteadyGray

Video input signals

DisplayPort 1.4 (2x)DisplayPort 1.4 (2x)DisplayPort 1.2 (2x)

DVI-D Dual Link,
DisplayPort

DVI-D Dual Link,
DisplayPort

DisplayPort 1.2 (x2)

DisplayPort 1.4 (x2)

DisplayPort 1.2 (2x)with built-in KVM switching

DisplayPort 1.2 (2x)SPECIFICATIONS
Warranty
Active screen size
(H x V)

Resolution

Luminance (DICOM
calibrated)
Contrast ratio

RapidFrame™, ULT, MXRT,
Intuitive Workflow Tools,
SteadyColor™, SteadyGray

SteadyColor, Color PPU, I-Luminate, RapidFrame, SmoothGray,
DuraLight, Clear/BlueBase, SoftGlow, Filmclip, MXRT, Intuitive
Workflow Tools with Touchpad

FDA: Medical Device Class II

FDA: Medical Device Class II

FDA: Medical Device Class II

FDA: Medical Device Class II

FDA: Medical Device Class II

FDA: Medical Device Class II

FDA: Medical Device Class II

FDA: Medical Device Class II

FDA: Medical Device Class II

Medical device
regulations

EC: Medical device class IIa(Regulation (EU) 2017/745

EC: Medical device class IIa(Regulation (EU) 2017/745

EC: Medical device class IIa(Regulation (EU) 2017/745

EC: Medical device class IIa(Regulation (EU) 2017/745

EC: Medical device class IIa(Regulation (EU) 2017/745

EC: Medical device class IIa(Regulation (EU) 2017/745

EC: Medical device class IIa(Regulation (EU) 2017/745

EC: Medical device class IIa(Regulation (EU) 2017/745

EC: Medical device class IIa(Regulation (EU) 2017/745

Package includes

Nio display system, MXRT
graphics controller, Intuitive
Workflow toolset, 5-year allinclusive warranty, QAWeb

Nio display system, MXRT
graphics controller, Intuitive
Workflow toolset, 5-year allinclusive warranty, QAWeb

Coronis Fusion diagnostic
display system, MXRT graphics
controller, Intuitive Workflow
toolset, 5-year all-inclusive
warranty, QAWeb

Nio display system, MXRT
graphics controller, Intuitive
Workflow toolset, 5-year allinclusive warranty, QAWeb

Nio display system, MXRT
graphics controller, Intuitive
Workflow toolset, 5-year allinclusive warranty, QAWeb

Coronis Fusion diagnostic display system, MXRT
graphics controller, Intuitive
Workflow toolset, 5-year allinclusive warranty, QAWeb

Nio display system including
built-in multimedia functionalities, MXRT graphics
controller, Intuitive Workflow
toolset, 5-year all-inclusive
warranty, QAWeb

Nio display system, MXRT
display controller, Intuitive
Workflow toolset, Warranty
for 5-years including 40,000
hour backlight, QAWeb

Coronis Uniti diagnostic
display system, MXRT graphics
controller, Intuitive Workflow
toolset, SoftGlow keyboard
and wall light, Touchpad andfilm clip

Options

Additional MXRT graphics
controller options, Additional
service and warranty options

Additional MXRT graphics
controller options, Additional
service and warranty options

Additional MXRT graphics
controller options, Additional
service and warranty options

Additional MXRT graphics
controller options, Additional
service and warranty options

Additional MXRT graphics
controller options, Additional
service and warranty options

Additional MXRT graphics
controller options, Additional
service and warranty options

Additional MXRT graphics
controller options, Additional
service and warranty options

Additional MXRT display
controller options, Additional
service and warranty options

Additional service andwarranty options

10

11

MDPC-8127
digital pathology

Digital pathology display

Meet MDPC-8127, Barco’s ultra-high definition 27-inch

MDPC-8127 specifications

medical display designed exclusively for digital pathology. With

Warranty

regulatory clearances for use in digital pathology including

Active screen size

569 x 335 mm (22.4 x 13.2")

Screen technology

IPS LCD with LED

primary diagnosis, it’s the first display that you can confidently
integrate into your digital pathology workflow with multiple
whole slide imaging systems.*

Watch the demo

*In the USA the MDPC-8127 can be used with WSI scanners and viewing
software that have been validated for use with the display.

Learn more

12

5 years

Diagonal size

684 mm (27”)

Aspect ratio

16:9”

Native resolution

8MP (3840 x 2160 @ 120Hz)

Luminance

450 cd/m² (max. 850 cd/m²)

Contrast ratio

1000:1

Color depth

10 bitColor gamut

132% sRGB

Sensor technology

I-Guard & ambient light sensors

Imaging technology

RapidFrame, PPU

Video input signals

2x DisplayPort 1.2

Medical device
regulations

FDA: Class II
EC: Medical device class IIa(Regulation (EU) 2017/745)

Package includes

MDPC-8127, QAWeb Enterprise,
Barco touchpad, 5-year warranty

Options

MXRT display controller
Intuitive Workflow Toolset
Additional warranty and service options

13

Eonisdentistry

Dentistry displays andclinical review displays

EMR

other

PIVOT

Ranging from 1MP to 8MP and available in different sizes, Barco’s Eonis clinical moni-

Eonis vs consumer displays

tors accommodate a wide variety of needs for viewing information and images, in thehospital and elsewhere. They combine consistent, high image quality and a versatile
design with centralized quality assurance, including a unique front sensor for consistent images.

Your benefits> Designed for hospital and dental environments, with various mounting options
and additional features

Watch video> Ambient Light pre-sets for dark rooms, offices and the OR> Supports efficient inter-disciplinary collaboration with consistent quality
controlled images> Cleanable displays for infection control (70% alcohol-based cleaning agents)> Sensor for bright and consistent image quality, and remote QA management
tools> Lowest cost-of-ownership over time without a need for capital budget> 3-year warranty

Models and options

14

15

D I S P L AY S

CLINICALD E N T I S T R Y D I S P L AY S

REVIEW

D I S P L AY S

MDRC-2222

MDRC-2324

MDNC-2123, option DE

MDNC-3521, option DE

MDRC-1419

MDRC-2421

MDRC-2422

MDRC-2324

MDRC-8132

Eonis 22” white

Eonis 24” white

Nio Color 2MP

Nio Color 3MP

Eonis 19”

Eonis 21”

Eonis 22”

Eonis 24”

Eonis Color 8MP

3 years

3 years

5 years

5 years

3 years

3 years

3 years

3 years

3 years

476 mm x 268 mm(18.7” x 10.5”)

518.4 mm x 324.0 mm(20.41” x 12.76”)

521 x 293 mm(20.5 x 11.5")

433 x 325 mm(17.1 x 12.8")376 x 301 mm(14.81 x 11.85”)

432 mm x 324 mm(17" x 12.76")

476 mm x 268 mm(18.7” x 10.5”)

518.4 mm x 324.0 mm(20.41” x 12.76”)

708.48 x 398.52 mm(27.9 x 15.7”)

Screentechnology

LCD

LCD

LCD

IPS-SFT Color LCD

IPS-TFT Color LCD

TFT Color LCD

TFT Color LCD IPS

TFT Color LCD IPS

IPSDiagonal size

546 mm (21.5")

609.6 mm (24.0”)

598 mm (23.6")541 mm (21.3")

482 mm (19”)

541 mm (21.3")

546 mm (21.5")

609.6 mm (24.0”)

812.80 mm (32.0”)

Aspect ratio

16:9

16:10

16:9

4:3

5:4

4:3

16:9

16:10

16:9

Native resolution

2MP (1920 x 1080)

2MP (1920 x 1200)

2MP (1920 x 1080)

3MP (1536 x 2048)

1MP (1280 x 1024)

2MP (1600 x 1200)

2MP (1920 x 1080)

2MP (1920 x 1200)

8MP (3840 x 2160)

Luminance(DICOM calibrated)

300 cd/m²(180 cd/m²)

Standard brightness version:
430 cd/m² (250 cd/m²)
High brightness version:
600 cd/m² (350 cd/m²)

370 cd/m²(460 cd/m²)

1050 cd/m²(600 cd/m²)

330 cd/m²(250 cd/m²)

440 cd/m²(250 cd/m²)

300 cd/m²(180 cd/m²)

Standard brightness version:
430 cd/m² (250 cd/m²)
High brightness version:
600 cd/m² (350 cd/m²)

500 cd/m²(300 cd/m²)

1000:1

1000:1

1000:1

2000:1

1000:1

1500:1

1000:1

1000:1

1000:1

Screen protection

Protective, anti-glare
glass coverProtective, anti-glare
glass coverOptional: protective cover,anti-reflective glass cover

Factory fit protective,
non-reflective glass coverOptional:
PCAP touchscreen

N/A

Optional:
PCAP touchscreen

Optional:
PCAP touchscreen

N/A

Front sensor
technology

Front consistency
sensor

Front consistency
sensorIntegrated front-of-screen
sensor, BLOS, ALC presets

Integrated I-Guard frontof-screen photometer
sensor, ALC

Front consistency
sensor

Front consistency
sensor

Front consistency
sensor

Front consistency
sensor

Front consistency
sensorVideo input signals

DisplayPort, DVI

DisplayPort, DVI

DisplayPort, DVI

DisplayPort 1.4 (2x)DisplayPort, DVI-I,
HDMITM

DisplayPort, DVI-I,
HDMITM

DisplayPort, DVI-I,
HDMITM

DisplayPort, DVI-I,
HDMITM

DisplayPort 1.4 (x2)

Medical device
regulations

FDA:
Medical device class I
EC: Medical deviceclass I (Regulation (EU)2017/745)

FDA:
Medical device class I
EC: Medical deviceclass I (Regulation (EU)2017/745)

FDA:
Medical Device Class II
C: Medical device classIIa (Regulation (EU)2017/745)

FDA:
Medical Device Class II
EC: Medical deviceclass IIa (Regulation (EU)2017/745)

FDA: Medical deviceclass I
EC: Medical deviceclass I (Regulation (EU)2017/745)

FDA: Medical deviceclass I
EC: Medical deviceclass I (Regulation (EU)2017/745)

FDA: Medical deviceclass I
EC: Medical deviceclass I (Regulation (EU)2017/745)

FDA: Medical deviceclass I
EC: Medical deviceclass I (Regulation (EU)2017/745)

FDA: Medical deviceclass I
EC: Medical deviceclass I (Regulation (EU)2017/745)

Package includes• Eonis display• 3-year all-inclusive
warranty• QAWeb• Eonis display• 3-year all-inclusive
warranty• QAWeb• Nio display• 5-year all-inclusive
warranty• QAWeb• Nio display system• MXRT graphics controller• Intuitive Workflow toolset• 5-year all-inclusive
warranty• QAWeb• Eonis display• 3-year all-inclusive
warranty• QAWeb• Eonis display• 3-year all-inclusive
warranty• QAWeb• Eonis display• 3-year all-inclusive
warranty• QAWeb• Eonis display• 3-year all-inclusive
warranty• QAWeb• Eonis display• 3-year all-inclusive
warranty• QAWeb• MXRT graphics controller• Intuitive Workflow
toolset• Versions: white/black,cover/no cover/touchscreen• Additional service andwarranty options• MXRT graphics controller• Intuitive Workflow
toolset• Versions: white/black,standard/high-brightness, cover/no cover/touchscreen• Additional service andwarranty options• Additional MXRT
graphics controller
options• Intuitive Workflow
toolset• Protective cover• Additional service andwarranty options• MXRT graphics controller• PCAP touchscreen• Intuitive Workflow
toolset• High-bright version• Additional service andwarranty options• MXRT graphics controller• PCAP touchscreen• Intuitive Workflow
toolset• High-bright version• Additional service andwarranty options• MXRT graphics controller• PCAP touchscreen• Intuitive Workflow
toolset• Additional service andwarranty options• MXRT graphics controller• PCAP touchscreen• Intuitive Workflow
toolset• High-bright version• Additional service andwarranty options• MXRT graphics controller• Intuitive Workflow
toolset• Additional service andwarranty options

SPECIFICATIONS
Warranty
Active screen size

Contrast ratio

Options• Additional MXRT
graphics controller
options• Additional service andwarranty options

The terms HDMI, HDMI High-Definition Multimedia Interface, HDMI Trade dress and the HDMI Logos are trademarks or registered trademarks of HDMI Licensing Administrator, Inc

16

17

MEDICAL ALL-IN-ONESSPECIFICATIONS

JAO 18 MkII*JAO 21*

MUIP-2213

2 years

2 years

3 years

a-Si TFT LCD

a-Si TFT LCD

a-Si TFT LCD

Diagonal size

18.5”

21.5"

13.3”

Aspect ratio

16:9

16:9

16:9

Native resolution

1366x768

1920x1080

1920x1080

Luminance

300 cd/m2

250 cd/m²

400 cd/m2

Touch screen

PCAP multi-touch with 10
fingers (supports gloves
touch)

PCAP multi-touch with 10
fingers (supports gloves
touch)

PCAP multi-touch with 10
fingers (supports gloves
touch)

Screen protection

N/A

N/A

Anti-glare treatment

Cleanability

IP65

IP65

IPX3

Video input signals

N/A

N/A

N/A

Medical Device
Regulations

Medical grade ITE device

Medical grade ITE device

Medical grade ITE device

Accessories

Extended Cables for ARS5
USB Cable
RJ11 6P6C Cable
RJ45 Cable
VESA Mounting Screws
Thumbscrew

Extended Cables for ARS5
USB Cable
RJ11 6P6C Cable
RJ45 Cable
VESA Mounting Screws
Thumbscrew

Desktop standJAO18 MKII,PSU 90W Medical Adapter (19V
@ 4.74 A Adapter)
EU Power cord

JAO21,PSU 90W Medical Adapter(19V @ 4.74 A Adapter)
EU Power cord,
Extended power cable

Warranty
Screen technology

Medical all-in-ones

Package includes

Options

MUIP-2213,
Sealed connector cover, AC/DCPower adapter (100-240VAC) &EU / US Power cord (dependingon the model)
User guide (English)

Ask about our different models and associated accessories*This product is subject to regional availability.

18

19

AMM & MDXC
surgery

Surgical andendoscopy displays

endoscopy

A family of surgical and endoscopy displays designed for real-time imaging in theoperating room. The portfolio comprises near-patient and large-screen displays for

4 ways 4K changes
surgical imaging:use in any environment where surgical procedures are carried out, including theendoscopy room, interventional suite or hybrid OR. Available in high definition or 4K
ultra-high definition.

Your benefits> Deep blacks and accurate colors from any viewing angle> 4:4:4 chroma subsampling for impeccable color accuracy> Near-zero latency provides perfect surgical hand-eye coordination> Improved visualization in high ambient light environments> Backlight output stabilization guarantees image stability and brightness overthe lifetime of the display

Download ebook> Cleanable infection control design complies with strict OR hygiene standards> Unique cable management system makes it easy to install and disinfect> Automated failover feature so a backup signal is always available> Perfect match with our Nexxis OR-over-IP platform for full OR integration

Models and options

20

21

SURGICALD I S P L AY S

E N D O S C O P Y D I S P L AY S

SPECIFICATIONS

L A R G E S C R E E N D I S P L AY S

AMM215WTTP

AMM240EDMDSC-8427 12G

MDSC-8527

MDSC-8231 12G

MDSC-8232 M3D

MDFC-8232 3xB

MDSC-8532

MDSC-8255

MDAC-8355

2 years

2 years

3 years

3 years

3 years

3 years

3 years

3 years

3 years

3 years

Active screen size
(H x V)476 x 267 mm(18.7 x 10.5”)

520.4 x 326 mm(20.46" x 12.83")

597 x 336 mm(23.5” x 13.2”)

596 x 335 mm(23.5” x 13.2”)

698 x 368 mm(27.48” x 14.49”)

698 x 368 mm(27.48” x 14.49”)

698 x 368 mm(27.48” x 14.49”)

708 x 399 mm(27.87" x 15.71")

1210 x 640 mm(47.64” x 25.20”)

1210 x 680 mm

Screentechnology

TFT AM LCD /LED Backlight

TFT AM LCD /LED Backlight

TFT AM LCD / IPS-PRO
technology / LED
backlight

TFT AM LCD /LED backlight

TFT AM LCD /IPS-PRO technology /LED backlight

TFT AM LCD /IPS-PRO technology /LED backlight

TFT AM LCD /IPS-PRO technology /LED backlight

TFT AM LCD / LED backlight

TFT AM LCD /IPS technology /LED backlighta-si TFT active matrix,
IPS

21.46” (545.2 mm)

24.07" (611.3 mm)

27" (685 mm)

27” (684 mm)

31” (789 mm)

31” (789 mm)

31” (789 mm)

32” (813 mm)

54.6” (1388 mm)

54.6” (1388 mm)

Aspect ratio

16:9

16:10

16:9

16:9

17:9

17:9/16:9

17:9/16:9

16:9

16:9

16:9

Resolution

Full HD1920 x 1080

WUXGA
1920 x 1200

4K UHD
3840 x 2160

4K UHD
3840 x 2160

4K-2K
4096 x 2160

4K-2K (4096 x 2160) /UHD (3840 x 2160)

4K-2K (4096 x 2160) /UHD (3840 x 2160)

4K UHD
3840 x 2160

4K UHD
3840 x 2160

4K UHD
3840 x 2160

Luminance

250 cd/m²

300 cd/m²

750 cd/m²

800 cd/m²

550 cd/m²

450 cd/m²

450 cd/m²

850 cd/m²

500 cd/m²

350 cd/m²

Contrast ratio

1000:1

1000:1

1400:1

1000:1

1400:1

1300:1

1300:11350:1

1000:1

1100:1

Front protection
screenpCap touch(10 points)

Anti-reflective PMMA
protective cover2-side anti-reflective
glass with anti-fingerprint

2-side anti-reflective
alkali-aluminosilicate
glass cover

Scratch-resistant 2-side
anti-reflective alkalialuminosilicate glass
cover

Scratch-resistant 2-side
anti-reflective alkali-aluminosilicate glass cover

Scratch-resistant 2-side
anti-reflective alkali-aluminosilicate glass cover2-side anti-reflective
alkali-aluminosilicate
glass cover

Scratch-resistant, 2-side
anti-reflective glass cover2-side anti-reflective
laminated safety glass

Cable management

n/a

Yes

Yes

Yes

Yes

Yes

Yes

Yes

Yes

N/A

Protection rating

IPx1

IP22

IP21
(IP45 front side)

IP21(IP45 front side)

IP20(IP45 front side only)

IP20(IP45 front side only)

IP20(IP45 front side only)

IP21(IP45 front side only)

IP20(IP45 front side)

IP20(IP45 front side)

Failover

N/A

N/A

Yes

Yes

Yes

Yes

Yes

Yes

Yes

Yes (black version only)

4K-UHD input, either:
Quad-link 3G-SDI,
12G-SDI, 1x DP 1.1,
1x DP 1.2 MST, 2x HDMITM
2.0 (all up to 4096 x 2160
@50Hz/60Hz) or
2x DP 1.1 (up to 2048 x
2160 @50Hz/60Hz)
FHD input (upscaled to
UHD):
1x DVI

2D mode:4K-UHD resolution:
3x DP 1.1, 1x DP 1.2 MST,
2x f.o. SFP + for Nexxis link
FHD resolution (upscaled to 4K):
1x DVI, 1x 3G-SDI
3D mode:4K-UHD resolution:
3x DP 1.1, 1x DP 1.2 MST,
2x f.o. SFP + for Nexxis link
FHD input:
1x DVI, 1x 3G-SDI

2D mode:4K-UHD resolution: 3x DP 1.1,
1x DP 1.2 MST, 1x 12G-SDI, 1x
HDMITM2.0 (3HB)
FHD resolution (upscaled to 4K):
1x DVI
3D mode:4K-UHD resolution:
3x DP 1.1, 1x DP 1.2 MST, 1x 12G-SDI(3SB), 1x HDMITM2.0 (3HB)
FHD resolution:
2x 3G-SDI (dual stream)

SSTF version
4K-UHD input:
1x DP 1.2 SST/MST
2x HDMITM 2.0 (all up to 3840 x
2160 @ 50/60 Hz)
FHD input (upscaled to UHD):
1x DVI, 1× 3G-SDI *;NXF version
4K-UHD input:
1x DP 1.2 SST / MST
1× HDMITM 2.0 (all up to 3840 x
2160 @ 50/60 Hz)
FHD input (upscaled to UHD):
1x DVI, 1x 3G-SDI
2xFO SFP + 4K-UHD Nexxis link*

2x DP 1.1, 1x DP 1.2 MST, 2x
HDMITM 2.0,
1x DVI, 1x 3G-SDI
Optional:
12G version:
2Quad-link SDI,
12G-SDI,integrated Nexxis version:
MNA-240 network adapter

2x DVI Dual link
4x DVI Single Link
1x DP 1.2 SST

2x 12G-SDI

2D mode:1x 3G-SDI, 1x DVI
3D mode:1x DVI(3G-SDI input loop through),

DP 1.2 SST, 3G-SDI

FDA: Medical Device Class I
EC: Medical Device Class I(MDR (EU) 2017/745)

FDA: Medical Device Class I
EC: Medical Device Class I(MDR (EU) 2017/745)

FDA: Medical Device Class I
EC: Medical Device Class I(MDR (EU) 2017/745)

Warranty

Active screen size
(diagonal)

Video inputs

Video outputs

Certifications

DVI-I, VGA(D-Sub, 15pin)

1xDVI-I, 1x HD15,
1x Y/C, 1x 3G-SDI, 1x
RGBS / YPbPr

4K-UHD input, either:
Quad-link 3G-SDI,
12G-SDI, 1x DP 1.1(all up to 4096 x 2160
@50Hz/60Hz) or
2x DP 1.1 (up to 2048 x
2160 @50Hz/60Hz)
FHD input (upscaled to
4K):
1x DVI-SL

SSTF version
4K-UHD input:
1x DP 1.2 SST/MST
2x HDMITM 2.0 (all up to 3840
x 2160 @ 50/60 Hz)

FHD input (upscaled to UHD):
1x DVI, 1× 3G-SDI*;NXF version
4K-UHD input:
1x DP 1.2 SST / MST
1× HDMITM 2.0 (all up to 3840
x 2160 @ 50/60 Hz)

FHD input(upscaled to UHD):

1x DVI, 1x 3G-SDI
2x FO SFP + 4K-UHD Nexxis
link*

2x 12G-SDI

SDI / S-video /Composite Video,
RGBS / YPbPr

1x DVI-I,
1x 3G-SDI

FDA: Medical Device
Class I
EC: Medical Device ClassI (MDR (EU) 2017/745)

FDA: Medical Device
Class I
EC: Medical Device ClassI (MDR (EU) 2017/745)(input loop through),

1x DVI(4K display screen clone downscaled to FHD)

FDA: Medical Device
Class I
EC: Medical Device ClassI (MDR (EU) 2017/745)

DP 1.2 SST,
3G-SDI
FDA: Medical Device
Class I
EC: Medical Device ClassI (MDR (EU) 2017/745)(input loop through),

1x DVI(4K display screen clone downscaled to FHD)

FDA: Medical Device
Class I
EC: Medical Device ClassI (MDR (EU) 2017/745)

The terms HDMI, HDMI High-Definition Multimedia Interface, HDMI Trade dress and the HDMI Logos are trademarks or registered trademarks of HDMI Licensing Administrator, Inc22

2x 3G-SDI
1x DVI

DVI (screen clone) 3G-SDI.
12G version:

12G-SDI
FDA: Medical Device Class I
EC: Medical Device Class I(MDR (EU) 2017/745)

N/A
Medical grade
ITE device* DP. HDMI and DVI all suport
HDCP 1.4

23

MDMX-MNA
surgery

In-wall displays for
the OR

Barco’s MDMX family is a series of in-wall consoles for the operating room. Combine
and view multiple images, by relying on an intuitive control system. The consoles
have built-in Nexxis functionality and a programmable Nexxis button, so they can be
integrated easily within both new and existing Nexxis systems.
Nexxis is available via OR solution providers, ask your contact person about it!

Your benefits> Enjoy ergonomic viewing support for everyone in the OR: from doctors andnurses to trainees> Place multiple images or video sources next to each other> Don’t lose space, the consoles are built into the OR wall
> Interact with remote-PC video sources as if you’re in front of them
> 2-year warranty

24

25

IN-WALL CONSOLES FOR THE OR

SPECIFICATIONSMDMX-22400 GNTBMDMX-25500 GNNBMDMX-22449 GNTB

2 years

2 years

2 years

Active screen size

15": 376 x 73 mm (14.8 x 2.9")24”: 527 x 296 mm (20.8 x 11.7”)

1210 mm x 680 mm(47.6 x 26.8")

15": 376 x 73 mm (14.8 x 2.9")24”: 527 x 296 mm (20.8 x 11.7”)
49”: 1074 × 604 mm (42.3 x
23.8”)

Screen technology

15": MVA/WLED
24": IPS/WLED

IPS/WLED15": MVA/WLED
24": IPS/WLED
49": IPS/WLED

Diagonal size

15": 384 mm (15.1")
24": 605 mm (23.8")

1388 mm(54.64")15": 384 mm (15.1")
24”: 605 mm (23.8”)
49”: 1232 mm (48.5”)

Aspect ratio

15": 160:31
24”: 16:9

16:9

15": 160:31
24”: 16:9
49”: 16;9

Native resolution

15": 1280 x 248
24": 1920 x 1080

3840 x 2160

15": 1280 x 248
24": 1920 x 1080
49": 3840 x 2160

Luminance

15": 300 cd/m²
24": 250 cd/m²

500 cd/m²

15": 300 cd/m²
24": 250 cd/m²
49": 500 cd/m2

Contrast ratio

15": 2000:1
24": 1000:1

1100:115": 2000:1
24": 1000:1
49": 1100:1

Protective, anti-reflective glass
front coverProtective,
non-reflective glass coverProtective,
non-reflective glass cover

N/A

N/A

N/A

1 x 10GbE Fiber Optic Interface for24” FHD Nexxis link
Optional: 2 x 10GbE Fiber Optic
Interface for 1x mobile device
connectivity on front2 x 10GbE Fiber Optic Interface for
55” UHD Nexxis link
Optional: 4 x 10GbE Fiber Optic
Interface for 2x mobile device
connectivity on front1 x 10GbE Fiber Optic Interface for24” FHD Nexxis link
2 x 10GbE Fiber Optic Interface for
49” UHD Nexxis link
Optional: 4 x 10GbE Fiber Optic
Interface for 2x mobile device
connectivity on front

Medical device
Class I

Medical device
Class I

Medical device
Class I• Touchscreen• Neutrik opticalCON accessories• Keyboard/trackpad module• Ask about our different model options
and associated accessories• Additional service and warranty
options• Neutrik opticalCON accessories• Keyboard/trackpad module• Ask about our different model options
and associated accessories• Additional service and warranty
options• Touchscreen• 1 or 2 keyboard/trackpad modules• Ask about our different model options
and associated accessories• Additional service and warranty
options

Warranty

Screen protection
Cleanability

Video input signals

Medical device
regulations
Options

26

27

Nexxissurgery

Platforms for thedigital operating room

Nexxis™ is our video-over-IP platform for the integrated digital operating room. It
makes it possible for you to share uncompressed, high-resolution video (and audio)in and between your operating rooms. This unique technology platform has been
specifically designed for integration into the digital OR.

Your benefits> Raw, uncompressed images – eliminating artifacts> Near-zero latency for perfect hand-eye coordination> 4K end-to-end> Simple to set up the OR (all you need is a single optical cable)> Easy plug and play of devices> Central & remote control> Always up to date with the latest networking and visualization technologies> Easy to adopt new integration requirements in the OR> Easy to connect new ORs to the network architecture> Expansion results in lower TCO

Nexxis WorkSpot
Control room solution
for interventional surgery
Offering one view of all information
as well as flexible layouts, Nexxis
WorkSpot has been designed to
improve teamwork and reduce
clutter and complexity in theinterventional control room. It
brings together all imaging sources
on a single display, managed by asingle keyboard and mouse.

Nexxis integrates into the majority of the world’s leading OR solution providers.
Ask your contact person about Nexxis, its benefits of operational efficacy and how
it can future-proof your ORs.

28

29

NexxisLive
Healthcare collaboration & conferencing solution

Remote work is everywhere, so why not make it happen in hospitals, too? NexxisLive
can be used by any healthcare professional - with any level of expertise; as well asCollaborate, educate anddemonstrate, live from your OR

medical device representatives or experts.
Basically, anyone in the surgical and interventional arena who needs to:> collaborate with remote users, to consult, to mentor or proctor sessions> assist a team during procedures or cases> teach to students who are in a university classroom or lecture hall
> broadcast a live-procedure to a congress venue – in real time under secured
bi-directional communication
With NexxisLive, you can talk, chat, share and draw as if you’re all in the sameroom. No matter how far away you are.

Barco technologies
that make a difference

Watch video

NexxisLive during surgery at AZ
Groeninge hospital, Belgium

Watch video

30

31

I-GuardTM front
sensor

Ambient Light
CompensationTM(ALC)

Uniform LuminanceTechnologyTM (ULT)

Color Per Pixel
UniformityTM (PPU)

FusionBarco Optical
GlassTM

This integrated, patented front-of-screen

Ensures that your diagnostic display

LCD panels typically suffer from non-Measures and adjusts the luminance of‘Fusion’ design replaces a dual-head

Reduces reflection, enhances image

photometer has become the industry-

system remains DICOM compliant underuniform behavior across more than 25%

each pixel and makes it permanently

display setup by a single, large-screen

sharpness and viewing comfort. The

standard technology for monitoring

all lighting conditions by monitoringof the screen area, causing noise in the

DICOM

compliant.

As

PPU

monitor without a hindering central

multi-coated glass is highly durable,

image quality and DICOM consistency.

display brightness compared to theimage. ULT eliminates those luminanceeliminates

screennoise,

ensuring

bezel. It allows you to place more images

scratch-resistant, and protects the LCD

The new generation of I-Guard has

ambient light level in the room. In

andconsistent colors and grayscales.

on the screen – just the way you want –panel from physical damage. A protective

been improved to measure much finercombination with QAWeb, ALC warns

sure that your display is compliant withwhich has proven to increase productivity

front cover – available on all Barco dis-

you when the ambient light level exceeds

DICOM GSDF across the entire screen

by up to 19% and reduce eye strain.

plays – further enhances the diagnostic

a predefined threshold.

area.

Benefits

Benefits+19%
gradationsof colors spread over thecolor gamut of the display. Combinedcolor

non-uniformities,

making

such,

+10%

experience.

with Barco’s QAWeb service, it provides
an easy-to-use tool for calibration andQuality Assurance.

Reduces eye strain

Figures originate from Montefiore study

Based on Montefiore study> Worry-free DICOM accuracy> DICOM compliance under all			> Continuous, automated QA of thelighting conditions

entire screen> Optimal image quality in every 		

entire optical stackenvironment so no detail goes 		> Remote assessment of yourunnoticed

display’s image quality

30% more compliance with ACR standards
for diagnostic
confidence> Uniform luminanceacross the 		> Perfect DICOM compliance from 		

Benefits

Benefits

Benefits> Reduced screen noise> Improved reading productivity> Protection of your valuable LCD 		> Improved contrast and accuracy 		> Easy side-by-side comparison of 		

10% more JNDs to unveil more subtle details
JNDs = Just Noticeable Differencesin fine details

studies> Improved image quality at a lower 		center to corner> Reduced eye fatigue

dose to the patient

+50%

+

FirePro GRAPHICS

Without I-Guard

With I-Guard

2x

mal reflections> Easy and safe cleaning+

screen real estate2024 to display
2025
images the way you wantpanel
> Improved image contrast & mini-		

FirePro GRAPHICS

Benefits
es productivity
by up to 19%

2026

50% more calibrated lightfor better detection

DICOM compliance under all lighting
conditions thanks to ALC

I-Guard video

100% uptime of your PACS system withintervention-free calibration and QA

Without ULTWith ULT

40,000ULT technology highlight

Only all-in-one display system, including
display controllers and software

Without PPUWith PPU

1stCoronis Fusion video

Technology highlight

Uniform luminancetechnology
Non-uniformity and noise in LCDs

2x the image detailwith SpotView Mag

Fewer image manipulations with Barco’s
tools for an intuitive workflow

Although LCD technology has improved a lot over the past
years, Liquid Crystal Displays still present a challenge whenit comes to showing images consistently over time. Today’s
LiquidCrystal Displays typically suffer from non-uniform
behavior across the screen area, causing arbitrary patterns
that interfere with the medical image content. These
patterns, also called mura or noise, can have several causes.
Non-uniformity of the backlight for example is responsible
for the typical luminance fall-off towards the corners of thedisplay. Also, liquid crystal cell artifacts can cause random,high frequency, pixel-by-pixel variations in the luminanceand color of the display. The superposition of all theseeffects results in spatial noise, which can be observed as afixed, cloudy pattern on the LCD.

Uniform image rendering across the entire
screenBacklight luminance warranty up
to 40,000 hours
To compensate for LCD non-uniformity, We
have has developed its proprietary Uniform
Luminance Technology. This technology
reduces luminance non-uniformities andcolor non-uniformities and makes sure thatthe display is compliant with DICOM GSDF across the entire
display area.
In a first step, Barco’s Uniform Luminance Technology
characterizes the spatial noise pattern of the display. Basedon this data, which includes all causes of non-uniformity,
appropriate correction values for every display pixel and all
video levels are calculated. The correction data is chosen
so that it will compensate for the spatial noise of the LCD
panel.

1st ever Fusion display on the market
and now its 5th generation

Uniform Luminance Technology will digitally precompensate in real-time each image that needs to be
displayed before sending it to the display panel. The
correction data that is applied is the inverse of the spatial
noise of the LCD. When this pre-compensated image isdisplayed, the stationary spatial noise of the LCD and thepre-correction cancel out. As a consequence, the resulting
image displayed is spatially.

Without Uniform Luminance Technology

Watch video

Spatial noise may have a negative influence on theaccuracy of reading medical images, as the presenceof noise makes it much harder for a human observer todiscriminate a (subtle) image feature from the surrounding
background. In other words: spatial noise increases the riskof “false negatives” (clinically relevant features that remain
undetected). A spatial noise pattern can also be confusedwith a clinically relevant image feature. The spatial noise
pattern of the display, for example, could be interpreted bya radiologist as actually being part of the medical image.
Such confusion can result into “false positives”.

Relative luminance (%)

Luminance non-uniformity

Watch video

With Uniform Luminance Technology

In the adjoining graphs, we see the luminance uniformity
for video level 64 (DDL 6.25%). Without Barco’s Uniform
Luminance Technology, luminance variations of over 40%
are possible, whereas with Uniform Luminance Technology,
luminance variations are reduced to less than 5%.

Relative luminance (%)

The clinical relevance of Uniform LuminanceTechnology

Download

32

33

IPS-Pro

RapidFrameTM

I-LuminateTM

SteadyColorTM &SteadyGrayTM

SpotViewTM

DimViewTM

A breakthrough LCD architecture offering

For high-speed cine imaging without

Simply push the I-Luminate™ ‘hot light’

Around 90% of images generated are

SpotView creates a bright focal point on

DimView automatically dims the auxiliary
displays used for patient worklists ora significantly wider viewing angle thanblur when reviewing multi-frame image

button to temporarily boost your dis-

now in color and color is often used forthe screen, enhancing brightness andcompeting LCD technologies. It ensuressequences – such as CT, tomosynthesis,

play brightness, and get a clearer view.

additional functionality (e.g. combining

contrast inside a circular region of inter-

dictation, reducing peripheral ambient

images are displayed more accurately andbreast ultrasound and breast MRI – or

Combined with Barco’s handy film clip,

Doppler images with ultrasound). In order

est while dimming the surrounding im-

light during readings so the eye can better

when hovering over the image with the

I-Luminate allows you to compare stud-to guarantee consistent, perceptually

age area. This heightens visibility of subtle

focus on details on the diagnostic display.

with a higher contrast ratio near the sidesof the display. With IPS-Pro, the viewing

magnifying glass. RapidFrame technology

linear color and still meet the DICOM

differences in the image. With SpotView

angle is 120°, which is nearly twice thecounteracts motion blur by more quickly

standard for grayscales, SteadyColor

Mag, you can even double the size ofvalue achieved by other technologies.

adapting pixels to a high frame rate.

executesforthe image so no detail goes unnoticed.

both grayscale and all Just Noticeable

SpotView has been proven to decrease

Differences (JNDs). This is achieved using

reading time by 15.5%, with 6% increased

multiple sensors in combination withreading accuracy in evaluation of mam-numerous 3D Look-Up Tables (LUTs).

mography contrast-detail phantom.

ies with film-based priors on the display.

advanced

SteadyGrayuses

calibration

the

same

color

calibration technology to achieve a moreconstant and consistent gray appearance.

Benefits

Benefits

Benefits

Benefits

Benefits

Benefits> High-quality visualization of images> High-speed cine imaging without 		

> Enhanced perception of small 		> Maximize the diagnostic value of 		> Better visibility of subtle differ		

> Improved focus on the main 		> Wider viewing angle improves 		multiuser viewing

blur for fast interpretation> Increased screening efficiency 		
due to highly efficient diagnostic 		
workflow

and low contrast details
based studies> Uninterrupted DICOM-compli-		

display, no distractions

ences in the image

color in images> Simplified comparison with film-		> Maintain image quality and> Extremely helpful when viewing 		

DICOM compliance under all 		

pediatric extremities, breast

lighting conditions

calcifications and chest exams> More ergonomic reading
experience

ance

IPS-PRO

1 JND

1 JND

1 JND

1 JND

1 JND

1 JND

1 JND

NON-IPS

SpotView

BeforeSteadyColor

RapidFrame video

SteadyColor technology
highlight

SpotView DimView

DimView Virtual View

Virtual View

After
SteadyColor

SpotView video

DimView video

AAM
SteadyColor & SteadyGray
Technology highlight

As the pioneer and industry leader of medical calibration technology,
Barco defined the image quality standard for diagnostic viewing
applications with the SteadyColorTM color calibration technique.
With our latest color display systems, QAWeb incorporates the vast
color know-how Barco has developed over the past several decades
into SteadyColor to further advance one of its core technologies.

Watch video

SteadyGrayTM uses the same color calibration technology to achievea more constant and consistent gray appearance. SteadyGray
ensures that all gray values are as close as possible to the tint of theselected white color, whether that is blue base, clear base, or someother preferred white tint.

Watch video

Watch video

Download

34

35

SmartCursorTM/
FindCursorTM

VirtualViewTM

Conference
CloneViewTM

Application
AppearanceManagerTM

User Profiles

SingleViewTM

The use of multiple resolution displays in

VirtualView provides radiologists with

Conference CloneView software is a

Application Appearance Manager allows

User Profiles creates a custom profile,SingleView enables the use of the entire

a PACS workstation results in the cursor

a virtual navigational head on the main

high-performance,

tool

you to individually manage the color

set per modality and per user. It enables

display as one display, and eliminates any

often getting stuck traveling from one

display. It appears and disappears on user

for projecting medical images from a

profile and luminance of each applicationa quick environment switch for image

tearing down the center of the display.

display to another, if you can even locate

command, allowing you to add or remove

diagnostic display onto a large screen,

window so content appears as intended.

type, such as a lower luminance forthe cursor in the first place. SmartCursor

this screen content at your convenience.

especially useful in classroom settings.

Benefits

Benefits

Benefits

Benefits

Benefits

Benefits> Avoid losing time locating or> View other screen content more 		> High-quality cloned image without 		

> Seamlessly integrate color (e.g. 		

> Read images according to your 		> More flexible desktop where im

X-ray images and hig

maps the cursor to the correct placesintheadjacent

display.

FindCursor

indicates the location of the cursor witha highlighted circle.

moving the mouse cursor

comfortably> Smoother and faster workflow> Ideal for viewing worklists, re		search, dictation, or ancillary tasksneeded; simplify the desktop

SmartCursor

SpotView SpotView

SmartCursor

Find Cursor

DimView DimView

Find Cursor

color photos) into the diagnostic 		losing pixel precision> Reduce the number of displays 		

oneVieww

easy-to-use

workflow> Fosters collaboration between 		peers
DimView
SpotView>SpotView
Ideal during multi-disciplinary		

DimViewof the screen
settings for every modality> More ergonomic viewing 			CloneView CloneView SmartCursor SmartCursor Find Cursor Find Cursor
experience

CloneView CloneView

SmartCursor
SmartCursor

DimView User Profiles User
VirtualView
Find CursorFind
CursorAAM
SpotViewScreenShot
AAM
ProfilesSoftGlow ScreenShot
SoftGlowSingleView

CloneView

Screenshot

SoftGlow

User Profiles

Capturing ideal images for meetings and

AAM

User Profiles SmartCursor / FindCursor video

SoftGlowon image content

team meetings

SingleView SingleView

ages can be placed in the center 		

> Automatically get the right display 		

> Manage the luminance of applica		
Virtual View Virtual View SingleView SingleViewtion windows depending 			

SoftGlow

Virtual ViewVirtual View

own preferences

VirtualView video

AAM

ScreenShotScreenShot

review, this tool enables you to take a
AAM
ScreenShot
screenshotof what is currently
in focus.

User Profiles
User Profiles

Appearance Manager video

SpotView

Benefits

User Profiles video

DimView

Virtual View

CloneView>SingleView
Easy to communicate and shareimages and data
> Fosters easy collaboration between
peers

Watch video
Watch video

Watch video

Watch video

AAM

ScreenShot

User Profile

Watch video36

37

Wide gamut & color
calibration

Composition,
PiP & PaP

Failover

BLOS (Backlight
Output Stabilization)

Nexxis Integration

Wide gamut and color calibration ensure

Composition,

Failover

Barco’s

Backlight Output Stabilization (BLOS)

true-to-life color reproduction essential&

PaP

Nexxis Integration allows Barco’s surgicalfeaturessurgical displays ensure uninterrupted

technology maintains optimal brightness

for accurate medical imaging. Barco’sinBarco’s

displays to seamlessly connect with theallow

performance by automatically switching

and contrast in surgical displays, ensuring

Nexxis digital operating room platform,

surgical displays provide consistent color

multiple image sources to be displayed

to a backup signal source in the event

consistent image quality and enhancing

providing uncompressed video, audio,

accuracy, enhancing visual clarity andsimultaneously,of a signal loss, maintaining continuous

the visibility of fine details.

and data transmission with near-zerodetail recognition, which is critical for

efficiency and providing comprehensive

surgical precision.

visualPiP(Picture-in-Picture)(Picture-and-Picture)
surgicaldisplaysenhancing

information

workflowduringcapabilitiesin

operation.

latency.

surgicalprocedures.

Benefits

Benefits

Benefits

Benefits

Benefits> True-to-life color reproduction> Simultaneous display of multiple image> Uninterrupted performance> Consistent image quality> Seamless connectivity> Enhanced patient safety> Enhanced detail visibility> Real-time collaboration> Continuous availability of critical> Reduced eye strain> High-quality, synchronized visual> Consistent color accuracy> Enhanced visual clarity and detailrecognition

sources> Enhanced workflow efficiency> Comprehensive visual information

visual information

information

FAILOVER

38

39

Icon overview

40Standard 5-year warranty

Front-of-screen sensor

DICOM grayscale andcolor

10 bit Look Up Table

Approved for medical
use

Color Per Pixel
Uniformity Correction

12 bit Look Up TableOn Screen Display

Screenshot of image
details

30 bit Look Up Table

Per Pixel Uniformity
correction

Application AppearanceManager

LED backlight

QA software included

A virtual display

Integrated Ambient Light
Compensation

Usable in portrait andlandscape mode

Highlights region ofinterest

Backlight Output
Stabilization

Touch screen interface

Dims auxiliary displays

Color display

Uniform Luminancetechnology

Locates the mouse
cursorGrayscale display

Exceptionally wideviewing angle

Avoids the cursor gettingstuck

Anti-reflective protective
front cover

Ultra-smooth grayscaleprecision

For large-screen
projection

Anti-reflective protective
front glass

Widescreen aspect ratio

4K native display
resolution of 4096 by
2160 pixels

Diagnostic Luminance(Ultra bright display)

DisplayPort interface

Allows daisy chainingof displays

Compliant with DICOM
specifications

Multiple built-in sensors

Displays a wide rangeof colors

Long-lifetime backlights

Cleared for breast
tomosynthesis

Glasses-based 3D
technology

Environmental-friendly
product‘Hot light’ button

Seamless wide-screen
desktop

High-speed cine imaging
without blur

Integrated front-ofscreen sensor

Front consistency sensor

PIVOT

Easily switch between 2
workstations attached tothe display
Dim and mask non-criticalareas for improved opticalaccuracy
Product has the Barcoecolabel41

About Barco Healthcare

Improving the quality and value of care are the leading priorities for healthcare
professionals today. . At a time when the amount of data and demand for mobility
are growing rapidly and quality and compliance regulations are becoming stricter,
healthcare systems struggle to work more efficiently, and to provide personalizedcare to a growing number of patients in an affordable way.
Focused on transforming the delivery of care, Barco connects healthcare professionals
at every patient touch point, from the imaging room, to radiology, through specialist
consultations and into the surgical suite. We offer a network of medical imaging
solutions that deliver the complete picture to support more informed decisions,when and where it matters most.
It’s why we are considered the gold standard for medical visualization and how weare there at every stage of the patient’s journey. So healthcare professionals can focus
on patients’ needs and, in the end, achieve the best clinical outcome.

Visioneering a bright tomorrow

1

Marchessoux, C., et al. (2011). Validation of a new digital breast tomosynthesis medical display.
Proceedings of SPIE, 7966, 79660R

2

Kimpe, T. R. & Xthona, A. (2012). Quantification of detection probability of microcalcifications atincreased display luminance levels. , Breast Imaging, Springer 7361, 490-497.

3

Mony Weschler, Montefiore Medical Center, 2012

M00193-R32-0724-PC July 2024
© 2023 Barco NV. All rights reserved. Reproduction in whole or in part without written permission is prohibited. All brand names and product names aretrademarks, registered trademarks or tradenames of their respective holders. Due to continued innovation, information and technical specifications are
subject to change without prior notice. Please check www.barco.com for the latest specifications.

www.barco.com/healthcarewww.barco.com/healthcare