®

+MOBIlab

Bluetooth 2.0 technolgy enables a wireless online-transmission of+biosignal data to the Pocket PC or notebook/PC with g.MOBIlab .

Sensors for g.MOBIlab+ :Transmission works well for distances of 30 meters (indoor) and up
to 200 meters (outdoor).

GSR, pulse, respiration, snoring
sensor, temperature sensor, ...
Ask for more sensors!

MOBILE & WIRELESS BIOSIGNAL ACQUISITION

Highlights+

Get rid of the cable ...

Mobile biosignal acquisition with Pocket PC or notebook / PCLog data ...
An integrated flash memory (Micro-SD card) enables data logging
even after the device is disconnected from the computer.

acquire EEG, ECG, EOG, EMG and other signals even outside your lab

With up to 2 GB space on the flash card a total recording time of
120 hours is provided.

on-line visualization and storage of up to 16 channels on a Pocket PC or notebook / PCvarious software solutions available (driver/API, recording software, MATLAB/SIMULINK/LabVIEW ...)
transmit online biosignal data wirelessly via Bluetooth 2.0 to a Pocket PC or notebook / PC

Start quickly ...

log data directly on an internal flash card memory (Mini-SD card)
The easy handling of hardware and software combined withintegrate the device into your real-time system under SIMULINK (BCI, neuro-, biofeedback)+The system is available as an 8-channel "EEG-version" and a "multipurpose version" with 6 amplifier inputs (4 channels for EEG/EOG
and 2 channels for ECG/EMG) and 2 analog input channels forexternal sensors. 8 digital channels can be used for trigger signals.
The device is battery-supplied and designed for long-term
operation. 16-Bit technology and 256 Hz sampling rate guarantee
high-quality biosignal recordings.

g.MOBIlab - g.tec’s mobile and wireless biosignal acquisition
system - is the perfect tool for recording multimodal biosignal
data on a standard Pocket PC or notebook / PC . It allows toinvestigate brain-, heart-, muscle-activity, eye movements,respiration, galvanic skin response and other body signals. Various
sensors and electrodes are available. Ask for complete packages
or customized solutions!

+highest signal quality makes the g.MOBIlab a favoured tool forscientists, teachers and system developers.
A large variety of available accessories and sensors enable the user
to investigate even unusual scientific questions.

Good connections ...
Various connector boxes for the different versions of the device
assure a safe and easy connection of electrodes and sensors.
Shielded cables reduce artifacts. Touch-proof safety connectors are
the standard. Ask for customized connectors and adapters if
required!

Technical details and specifications
Colorful solutions ...

g.MOBIlab+(8-channel EEG version)

EEG Channels: 8, Filters: 0.5 - 100 Hz, Sensitivity: 500 µV(monopolar), no additional analog inputs

g.MOBIlab+(multi-purpose version)

EEG Channels: 2
Filters: 0.5 - 100 Hz
Sensitivity: 500 µV(bipolar)

Analog inputs (multi-purpose)

Channels: 2, Filters: DC-100 Hz, Sensitivity: +/- 250 mV (monopolar)

Additional inputs/outputs

4 digital inputs/outputs, 4 digital inputs (TTL), +5V

Look up our website

Power supply

4 standard AA batteries or accumulators (25 - 100 hours operation time, depending on mode)or contact us at any time toData acquisition

ADC with 16 Bit and 256 Hz/channel, serial interface (RS232), Bluetooth 2.0 / class I (+)

find out more about available software
solutions and how to use g.MOBIlab+Standard

Manufactured according to IEC 60601-1, for research application, no medical device

Internal storage card

Micro-SD flash memory card (up to 2 GB), accessible via the battery compartment

Weight360 gram (including batteries)

Dimension

155 mm x 100 mm x 40 mm

EEG/EOG Channels: 2
Filters: 0.01 - 100 Hz
Sensitivity: 2 mV(bipolar)

For ECG and EMG recordings a patient cable for use with adhesive
disposal Ag/AgCl electrodes is provided.

ECG/EMG Channels: 2
Filters: 0.5 - 100 HzSensitivity: 5mV(bipolar)

Acquire biosignals reliably even in sportsmen, vehicle drivers,
pilots or small children.

Perfect timing ...

in MATLAB/SIMULINK or LabVIEW!

With up to 8 digital I/O signals even tricky experimental paradigms
can be realized. All triggers are recorded synchronously with thebiosignals.
Simple pushbuttons allow the subject to respond to experimental
stimulation or to mark events during recording.

8-channel active EEG system
8-channel EEG version
enabledInstead of the standard EEG electrode connector box g.tec's active electrode system"g.GAMMAsys" can be used. No skin preparation and no abrasive gel is required anylonger. High quality EEG recording with a fully wearable active system!

multi-purpose version

g.tec is an official MATLAB partner.
(The MathWorks, Inc., Natick, MA.)
Application example: Navigation in the virtual reality with a BCI
MATLAB® and SIMULINK®
are registered trademarksof the MathWorks Inc.

A Brain-Computer Interface (BCI) is used to navigate in a VR environment and tocontrol simulated installations. EEG data are sent wirelessly to the computer so thesubject can freely move around. See also g.tec's virtual reality system "g.VRsys"!