Prime 95B 22mm™ Scientific CMOS Camera Datasheet
Scientific CMOS, EMCCD and CCD Cameras

High Resolution, High Speed
Global Shutter sCMOS
Teledyne Photometrics puts the right tool in your hands when
the goal is fast, high resolution imaging and documentation intrue plug-and play fashion. This 7 MP global shutter sCMOS
camera provides a large field of view and small pixels to maintainhigh resolution imaging at low magnification.
At 50 frames per second with a true global shutter, it is capableof high speed imaging without the possibility of distortion when
imaging fast moving objects. Even with this high speed andlarge number of pixels, the camera handles the data smoothly
via the new super speed USB 3.2 Gen 2 interface, which makes
integration and connectivity simple

73% Quantum Efficiency
4.5μm x 4.5μm Pixel Area
50 fps Full-Frame• Easy to integrate• Simple connection• High framerate• True global shutter

Global Shutter sCMOS
USB 3.2 Gen 2 Interface

Specifications

Camera Performance

SensorPhotometrics Sony-based Scientific CMOS

Active Array Size

3200 x 2200 (7 Megapixel)

Pixel Area

4.5µm x 4.5µmSensor Area

14.4mm x 9.9mm, 17.5mm diagonal

Peak QE%

73%

Read Noise

2.2 e-

Full-Well Capacity

8200e-

Framerate (full-frame)

50 fpsBit Depth

12-bitShutter Type

Global Shutter

Cooling

UncooledInterface

USB 3.2 Gen 2 (10 Gbps)

Triggers

1 Input
1 Output

High Speed Imaging

Scientific CMOS Camera Datasheet

Frame Rate

Accessories (Included)

Array Size

Speed

3200 x 2200

51

3200 x 1100

100

3200 x 550

1933200 x 275
3200 x 8

A00

REVA00837DATEAPPROVEDDLEISTER

2019-12-26

456A00

REVA00D3

REVISIONSDESCRIPTIONPRELIMINARYECO#SENSOR SCANNING
VERTICAL DIRECTION
8427671-635-005

501018.03 (SENSOR PLANE)CSENSOR SCANNING
VERTICAL DIRECTION
16.33 (2ND WINDOW EXT. SURFACE)

D42

USB 3.2 GEN210Gbps

405010A(0,0) PIXEL

4X 13

SENSOR SCANNING
HORIZONTAL DIRECTION

4X M3X0.5

I/O

C

44X 13A
18.03 (SENSOR PLANE)

I/O

A00

A

13.85 (1ST WINDOW EXT. SURFACE)

16.33 (2ND WINDOW EXT. SURFACE)

13.85 (1ST WINDOW EXT. SURFACE)

0 (C-MOUNT FLANGE)

1REVISIONSDESCRIPTIONPRELIMINARY

DC22071

ECO#71-635-005

3.2 GEN210Gbps

Trigger Cable

0 (C-MOUNT FLANGE)

5-005

Quickstart Guide353
456USB 3.2 CableB

USB 3.2 GEN210Gbps
A(0,0) PIXELSENSOR SCANNING
HORIZONTAL DIRECTIONB

4X M3X0.5

SECTION A-A (CROPPED)SCALE 2 : 1A

• DIMENSIONS ARE IN MMPART VERSIONTITLEALL INFORMATION CONTAINED HEREIN IS THE PROPERTY OF
TELEDYNE QIMAGING/TELEDYNE PHOTOMETRICS AND ISCONSIDERED PROPRIETARY AND SUBJECT TO RETURN UPONDEMAND. ACCEPTANCE AND RETENTION HEREOF SIGNIFIESAGREEMENT BY THE RECIPIENT THAT THE INFORMATION WILLNOT BE DISCLOSED TO OTHERS NOR USED CONTRARY TO THE
INTERESTS OF PM/QI WITHOUT PRIOR PERMISSION.

m.
1.5, THICKNESS 0.5 mm.

SECTION A-A (CROPPED)SCALE 2 : 1Teledyne
Photometrics is a registeredof Teledyne Photometrics. 3465 trademark. Kinetix is a trademark
All other brand and product names are the trademarks of their respective owners.

1.
2.
3.

24

CONFIDENTIAL

B, EXHAUST ON ALL FOUR SIDES.

I/O

SCALE: 1:1UNLESS OTHERWISE SPECIFIED:A

4SIZE

B

SPEC;ENGR;DIM;ALTA
DWG. NO.

71-635-005

21UNLESS OTHERWISE SPEC• DIMENSIONS ARE IN MM

ALL DIMENSIONS ARE REFERENCE.

CONF

Specifications
thisREAR
datasheetare subject
to change.EXHAUSTRefer to theONTeledyne
Photometrics
AIR INTAKE inON
FACE
OF CAMERA,
ALL FOURSIDES.website for most current specifications.

WINDOW SPECIFICATIONS
1.
1ST WINDOW:
HK9, THICKNESS 1 mm.
www.photometrics.com
2.
2ND WINDOW: REFRACTIVE INDEX 1.5, THICKNESS 0.5 mm.

8
Rev A2-12072020

photometrics.info@teledyne.com/ tel: +1 520.889.9933
76AREV.

A00ALL SHEETS ARE THE SAME REVISION STATUS SHEET 1 OF 15ALL INFORMATION CONTAINTELEDYNE QIMAGING/TELEDCONSIDERED PROPRIETARY
DEMAND. ACCEPTANCE A
AGREEMENT BY THE RECIPIENOT BE DISCLOSED TO OTHINTERESTS OF PM/QI WITHOU

43