• SONET/SDH Systems• Optical Communication• Pre-amp, Booster & In-line emulation• CATV• Laboratory

Description

The compact turnkey benchtop or 19”
rackmount unit incorporates a user-friendly
front panel housing with a LCD monitor
display, key switch, power control knob andoptical connectors. An integrated RS232
computer interface enables easy control,diagnostic functions and data acquisition.

AEDFA-PA-35-B-FA Gain
6050GaindB)

Amonics’ EDFA range adopts uniquedesign to produce maximum signal gain andsaturated output power while maintaining
low noise figure, enabling test capabilities insystem or component level manufacturing
and characterizing processes, as well as
facilitating highly demanding R&D
applications.

4030
20101530

1536

1542

1548

1554

1560

Wavelength (nm)
-10 dBm-15 dBm-20 dBm-30 dBm-35dBm-40dBm-25 dBm

AEDFA-PA-35-B-FA Noise Figure54.5

Noise Figure (dB)Pre-Amp, Booster and In-line Amplifier

Applications

43.5
32.5

Key Features• Turnkey device• RS232/Ethernet computer interface• High output power• High gain
• Low noise figure• Long operating life21530

1536

1542

1548

1554

1560

Wavelength (nm)
-10 dBm-15 dBm-20 dBm-30 dBm-35dBm-40dBmThe product is manufactured under a HKQAA ISO 9001 certified quality management system
The ISO 9001:2008 certification applies to the Hong Kong production site only

-25 dBm

Pre-amp Specifications
AEDFA-PA-30

Pre-Amp, Booster and In-line Amplifier

Input Signal Level
Optical Gain @ -40dBm>30dB

Optical Gain @ -10dBm

>20dB

>35dB
>30dB

Noise Figure @ -20dBm< 4.3dB

Operation Wavelength

1530nm to 1563nm

Input / Output IsolationPolarization Dependent Gain
Polarization Mode Dispersion

General Parameters

AEDFA-PA-35-40 to -10dBm> 30dBTyp. 0.3dB, Max. 0.5dB< 0.3psOperation Temperature : 0 to 40 oC
Storage Temperature:-10 to 70 oC
Power Supply (VAC): 90 – 240, 47 – 63Hz
Dimensions(mm): Benchtop: 260(W) x 330(D) x 120(H)
1U Rackmount: 485(W) x 360(D) x 45(H)
Other standard rackmount sizes are also available
Safety Control: Key-lock switch, BNC interlock key
Power Monitoring: Output power, Input power (optional)
Remote Control Port: DB-9 female (RS232)
RJ-45 (TCPIP/Ethernet) (optional)
Protection:
Pump laser (TEC) overheat
Optical Fiber:
SMF-28

Booster Specifications
AEDFA-BO-13
Input Signal Level

AEDFA-BO-18

AEDFA-BO-23-6 to +3dBm

Saturated Output Power@-3dBm input Signal> +13dBm> +18dBm> +23dBm

Noise Figure@-3dBm
Input Signal< 5.0dB< 5.5dB< 6.0dB

Optical Gain@-6dBm
Input Signal> 19dB> 24dB> 29dB

Operation Wavelength

1530 nm to 1563nm

Input / Output Isolation> 30dB

Polarization Dependent Gain

Typ. 0.3dB, Max. 0.5dB

Polarization Mode Dispersion< 0.3ps

In-line Specifications
AEDFA-IL-13
Input Signal Level

AEDFA-IL-18

AEDFA-IL-23-25 to -3dBm

Saturated Output Power@-3dBm input Signal> +13dBm> +18dBm> +23dBm

Noise Figure@-20dBm
Input Signal
Optical Gain@-25dBm
Input Signal
Operation Wavelength< 5.0dB< 5.5dB< 6.0dB> 25dB> 30dB> 35dB

Input / Output Isolation
Polarization Dependent Gain
Polarization Mode Dispersion

1530nm to 1563nm> 30dBTyp. 0.3dB, Max. 0.5dB< 0.3psOrdering Information
Product CodeAEDFA-aa-bb-c-dd

aa: PA for Pre-amp, BO for Booster, IL for In-line
bb: Saturated output power in dBm
c: B for Benchtop case, R for 1U 19”Rackmount case
dd: FA for FC/APC, FC for FC/UPC
SA for SC/APC, SC for SC/UPC

Amonics undertakes continuous and intensive product development to ensure its product performance at the highest technical standards. As a result,the specifications in this document are subject to change without notice.
Amonics Limited. 14/F, Lee King Industrial Building, 12 Ng Fong Street, San Po Kong, Kowloon, Hong Kong
Beijing Amonics Co. Ltd. Room 902, Unit 1, No.99 Chaoyang North Road, Beijing China 100025
Email: contact@amonics.com
HK Tel: +852 2428 9723

Website: www.amonics.com
HK Fax: +852 2428 9704

Beijing Tel: +86 10 84783386

Beijing Fax: +86 10 84783396