Optical Sensor

4NBSU-"$A contactless angle measurement device by laser

5JMU.FBTVSFNFOU

5JMUBOHMFBEKVTUNFOU

5JMUӖInstant non-contact angle measurement-BTFS"VUPDPMMJNBUPS

ӖWide working distance(Angle Range ±0.5deg at W.D.300mm)
ӖHigh precision and resolution
ӖSingle and multiple sport measurement mode available
ӖRed, Green, Blue and Infrared laser available
ӖExternal beam measurement available
Note : Select Single or Multiple spot mode when ordering
All wavelength(RGB,IR) system can be measured External beam.

<Measurement comparison>
Conventional way

Laser Autocollimator method

ΖLaser displacement sensor method
Displacement
sensor

Displacement
sensor<Solutions>
٬Easy measurement witholy one sensor, even for
hard to reach and microobjects.

<Topic>٬With tow sensors, hard toreach and measure microobjects.
)JHI4QFFEBOE
)JHI3FTPMVUJPO(FOFSBM
1VSQPTF

Calculate the Angle from between two point.
[ RUN]
233

ΗOptical Autocollimator method"DDFTTPSJFT

OK
L90 ９
０
*X

MonitorX = +2' 06"
Y = - 0' 51"
d = 2' 31"<Topic>٬Visual measurement

Monitor٬Expensive

Optical Autocollimator<Solutions>
٬Digital measurement
٬Competitive price

Easy tilting angle measurement for various DUT(Device Under Test)
Leaser beam needed for measurement

5005

Laser Autocollimator

5JMU.FBTVSFNFOU

ӖBasic composition of Smart LAC series
ΖSensor Head H410 Series
ΗSenor Cable HSHC1 Series
ΘImage Processing Unit HPU Series-BTFS"VUPDPMMJNBUPSΛ Tilt Stage

ΖSensor HeadΘImage Processing UnitIPU

ΚMonitor

Μ Stand Unit Set
ΗSensor CableΝ MirrorΞLAN CableΙAC Adapter
12Pin DIO Cable
Prepare by UserӖBasic Set
Standard Configuration
For Single Spot Measurement
Sensor Head Model[ΖSensor HeadΘIPU]

Note 1For Multiple Spot Measurement
ΘIPU

Wave Length

Model

Model
HPU-1000ΗSensor Cable

Model

Wavelength

Model

HPU-500SET-175R

655nm±10nm

HSHC1-1.5 (1.5m)

HPU-500SET-175S1

852nm±10nm

HSHC1-4 (4m)

HPU-500SET-175G1

520nm±10nm

HSHC1-10 (10m)

HPU-500SET-175B1

ΗSensor Cable

Model

ΖSensor Head

450nm±10nm

H410-175R

655nm±10nm

HSHC1-1.5 (1.5m)

H410-175S1

852nm±10nm

HSHC1-4 (4m)

H410-175G1

520nm±10nm

HSHC1-10 (10m)

H410-175B1

450nm±10nm

Note1:For Single spot measurement, Sensor Head and IPU are sold as a set.

)JHI4QFFEBOE
)JHI3FTPMVUJPO(FOFSBM
1VSQPTF"DDFTTPSJFT

ӖOptionStandard Configuration
ΙAC Adapter

ΚMonitor

ΛTilt Stage

ΜStand Unit Set

ΝMirror

Model

Model

Model

Model

Model

Model

Parallel Mirror:
HS-0(0deg)
Wedge Mirror:
HS-025AL(0.25deg)
Wedge Mirror:
HS-050AL(0.5deg)
Wedge Mirror:
HS-100AL(1.0deg)

RS232C Cable:
HRSCC1-2
LAN Cable:
HLANC1-2

HDC24V-2710MA

HMNT1

HB10

HA14

ΞLAN Cable

5006

Optical Sensor

New

Smart LAC Series Sensor Head : H410

5JMU.FBTVSFNFOU

ӖFeatureӖLaser Class 1
Can be used at any working environment (former Laser Class isClass2)
ӖHigh Anti-vibration performance Max.2G(X/Y/Z Axial direction)
ӖHigh performance with Light and Small size

-BTFS
"VUPDPMMJNBUPSӖApplications

Mirror Angle measurement for Camera device

Measure Glass Parallelism
Model
WavelengthSPEC
H410-175S1
H410-175G1
852nm±10nm
520nm±10nm
0.78mW
0.39mW
Class 1
Class 1
±0.5deg :0֐300mm±1.0deg :0֐200mm±1.75deg :0֐120mm
Appr. Į1.0mm
1sec

H410-175R
655nm±10nm
0.39mW
Class 1

Laser Power
Measurement are
by Working DistanceLaser Spot Size
Repeatability(Ave.)±0.2% of Full Scale(W.D.=200, Full Scale=±1deg)

Linearity(Ave.)±0.4% of Full Scale(W.D.=200, Full Scale=±1deg)

Sampling Speed
Calibration data
WeightAnti-vibration)JHI4QFFEBOE
)JHI3FTPMVUJPO(FOFSBM
1VSQPTF

25msecStored in each sensor head160g
Max 2G (X/Y/Z axial direction at 10֐500Hz)

The sensor head cannot be used alone. A combination with an image processing unit is required.
This product does not require initial calibration.
There are no restrictions on the combination of the sensor head and image processing unit.

ӖDimensions

58

2.5

2.5

2-ĕ3.529"DDFTTPSJFT

3-M3Depth4.5

20
11
15

42.7

40.2

11

11

ĕ18

64
68.2

21
40

5

007

2.5

35

10

50

3-M3Depth5

H410-175B1
450nm±10nm
0.39mW
Class 1

New

Smart LAC Image Processing Unit :HPU

.VMUJQMF
4QPU
.FBTVSFNFOU

EJFSFOU$FOUFS
PG(SBWJUZ
BOBMZTJT; PPN  

٬٬٬
TFSJBM$PNNVOJDBUJPOBWBJMBCMF*0$POUSPMMBCMF

ӖSingle-spot and Multiple-spot measurement are selectable

5JMU.FBTVSFNFOUӖMultiple-spot measurement
٬Max 5spot measurement simultaneously-BTFS"VUPDPMMJNBUPS

٬Sorting by Spots angle or area
٬Measure both Spots angle and relative angle between each spots
ӖSingle-spot measurement
٬Select one spot to measure from Max 5 spots.
ӖInternal Laser and External Laser mode available*Note: Please specify the wavelength and single-spot measurement /multi-spot
measurement mode when ordering.

Various communication capability can allow to operate Laser Autocollimator through Touch panel, PC or PLC.
For standalone usage, connect to either Touch panel or PC to control Laser Autocollimator.
For Embedded system, Connect to either PLC or PC to control Laser Autocollimator.
*:For PLC and PC usage, a Control software is needed to prepare by end user side.

IPU(Imagining Precess Unit)

PC

Sensor Head

RS232C or Ethernet communication enables
control and reading of measured values froma PC.
Ethernet Cable)JHI4QFFEBOE
)JHI3FTPMVUJPO(FOFSBM
1VSQPTF

RS232C Cross Cable
Sensor Cable"DDFTTPSJFT
USB Cable for Touch Panel
I/O Cable

MonitorMonitor power supply cable

PLC(Programmable Logic Controller)

MonitorDC 24V Power supply

5008

Optical Sensor

New

Smart LAC Image Processing Unit :HPU

5JMU
.FBTVSFNFOU

Function

HPU-1000

HPU-500

Single-spot measurement

ջջMultiple-spot measurement

Max 5 Spots‫٭‬Center of Areaջջ

Luminance centroidջջLuminance peakջջLuminance value

ջ

ջ

Zoom

ջ

ջ

Image rotation and reverseջջCircleջջRectangleջջOffset judgementջջSpecified spotջջAll spotsջ‫٭‬

Input/Output

ջջInput1 Hold resultsջջ

Input2 Zero resetջջ

Input3 Start trigger (24V)ջջ

Input4 Laser power ONջջ

Input5 Start trigger (5V)ջ

ջ

Output1 Judgement strobe signal

ջջOutput2 Resultsջջ

Output3 Busy for triggerջջAngle Measurement

Angle Analysis-BTFS"VUPDPMMJNBUPSConvenient Functions

Pass/Fail Judgment
Judgement Area

Pass/Fail Judgment
Judgement Conditions

Digital I/O)JHI4QFFEBOE
)JHI3FTPMVUJPO(FOFSBM
1VSQPTF"DDFTTPSJFT

5

009

New

Ӗ%JNFOTJPOT5JMU.FBTVSFNFOU①
Function
Power Lamp
Ethernet connecter
Sensor cable connecter
USB connecter for Touch Panel
Monitor connecter
Power supply for monitor
RS232C connecter
12-pole terminal block
DC24V terminal block

②

⑦
-BTFS
"VUPDPMMJNBUPS

③
⑧

④
⑤

⑨

⑥

134.4

60

)JHI4QFFEBOE
)JHI3FTPMVUJPO(FOFSBM
1VSQPTF"DDFTTPSJFT

167

4-M4Depth7

88

18

9.5

126

41

No.
Ζ
Η
Θ
Ι
Κ
Λ
Μ
Ν
Ξ

5

010

Optical Sensor

New

Smart LAC Series Software : HPU
ӖVarious center of gravity analysis

5JMU.FBTVSFNFOU

Three types of spot center-of-gravity analysis modes are available for more accurate measurements.
The center of area is suitable for specular reflectors with high flatness precision such as mirrors and glass, and the center of luminance is suitable forcases where reflection is distorted, such as the edge of a lens.
ӖMeasurement for Center of Area

-BTFS
"VUPDPMMJNBUPSBIN"SFBDFOUSPJE

5ISFTIPME

ӖMeasurement for Center of Luminance GRAY-VNJOBODF
DFOUSPJEPG
UIFBSFB-VNJOBODF1FBL

5ISFTIPME

1PTJUJPOPGUIFDFOUFSPGHSBWJUZ.JSSPSBOEHMBTTӖLuminance Peak PEAK

5ISFTIPME

1PTJUJPOPGUIFDFOUFSPGHSBWJUZ

1PTJUJPOPGUIFDFOUFSPGHSBWJUZ&EHFTVSGBDFPGSFTJO-FOT

ӖSingle-spot measurement
Single-spot measurement is a method of measuring the reflected light from a specified point out of multiple reflected lights.

Single-spot measurement SINGLE)JHI4QFFEBOE
)JHI3FTPMVUJPO(FOFSBM
1VSQPTF"DDFTTPSJFT
Angle measurement by specifying any one
point from a maximum of 5 light points

ӖMultiple-spot measurement
Multiple-spot measurement can be selected from two types of measurement according to the purpose.
մմմMultiple A: Measure each spot angle simultaneouslyմմմMultiple B: Measure relative angle between each spot simultaneously

5011

Multiple-spot measurement MULTI A

Multiple relative measurement MULTI B

Each spot angle measurement simultaneously

Relative angle measurement between eachspot simultaneously

NewӖSize orderӖAngle order
Center of gravity
analysisCenter of Area
(BIN)

Center of
Luminance(GRAY)
Identify spots size automatically and label
them in increasing order.

Identify spots angle automatically andlabel them in increasing order.

Peak Top(PEAK)

Sorting

Single Spot
Measurement

Multiple Spot
Measurement

ջ

ջ

ջ

ջ

ջ

ջ

ջ

ջ

ջ

Not available

Angle
Ascending order
Size
Descending order
Angle
Ascending order
Size
Descending orderMax. Luminance point

5JMU.FBTVSFNFOU

ӖLabeling Function
Each spot can be labeled when measuring multiple reflectors. Select a mode for labeling by size order or labeling by angle order.

-BTFS"VUPDPMMJNBUPS

ӖConvenient functions
ӖFunction 1 Image rotate
X-Y axial coordinate is not changed

Image can be rotated without rotating sensor head (±90deg)

ӖFunction 2 Flip image
X-Y axial coordinate is not changed

8JUIPVUDIBOHJOHTFOTPSNPVOUJOHQPTJUJPOPSBOHMF UIFWJFXJOHJNBHFDBOCFDIBOHFECZUIFGVODUJPOImage can be inverted without mounting sensor position

ӖFunction 3 Off set origin point)JHI4QFFEBOE
)JHI3FTPMVUJPO(FOFSBM
1VSQPTF"DDFTTPSJFT

Origin point can be off set to any pointӖZOOM (x2,X4,X8)

When the spot move into specified magnified area, viewing image is enlarged automatically.

ӖExternal communication port

The controller can be operated through Ethernet, RS232C and I/O

5012

Optical Sensor

New

Smart LAC Series Specification

5JMU.FBTVSFNFOUNOTEPlease note that part of the specifications of this product may be changed without prior notice in order to improve its performance.

Sensor Head
Item

-BTFS
"VUPDPMMJNBUPS

Model

SPEC
H410-175R

Wavelength
Power
Laser

655nm±10nm
0.39mW
Class 1

Output position
Out going angle

520nm±10nm

0.78mW
0.39mW
Class 1
Class 120±0.5mm at Reference surface 1
15±0.5mm at Reference surface 2

Max 5mW : Note1

Incident position

Within Radius 7.5mm from center of lens

Incident angle

Within ±3.5deg from center axial of lens.

Laser Spot size

Appr. Į1mm

Indicated resolution

1 sec

Repeatability(Ave.)

1 sec±0.2% of Full Scale(W.D.=200mm,
Full Scale ±1.0deg)

Linearity(Ave.)±0.4% of Full Scale(W.D.=200mm, Full Scale ±1.0deg)

Sampling speed

25mSec ֐

Operation display

LED(Green/Red)(FOFSBM
1VSQPTF"DDFTTPSJFT

852nm±10nm±0.05deg

Power)JHI4QFFEBOE
)JHI3FTPMVUJPO

H410-175G1

Environment

WeightAmbient temp. foruse
Ambient humid. foruse
Performance
guarantee
temperature

0֐40deg
35֐85% RH
23±5degstorage Temp.

-10֐+60deg

Anti-vibration

Max 2G (X/Y/Z axial direction) at 10֐500Hz
Appr. 160g

Note 1: Depending on Wavelength, beam size and power, ND filter may be needed for use.

5013

H410-175B1±0.50deg at W.D. 0֐300mm±1.00deg at W.D. 0֐200mm±1.75deg at W.D. 0֐120mm

Specifications

External Laser

H410-175S1

450nm±10nm
0.39mW
Class 1

New

Image Processing Unit
SPEC

Model

HPU-500 (Note 1),HPU-1000

Adaptable number of sensor head1 unitApplicable Sensor Model

H410-175R, H410-175S1, H410-175G1, H410-175B1

VoltageRating Power

Current
OutputPin assign

12-poleconnecter

Output circuitfor 2,3,4

Input circuitfor 2,3,4

Input circuitfor 5(+) and 6(-)

Input circuitfor 7(+) and 8(-)

Monitor output

Serial input

Weight1.8A or lessSensor Head:12V, 0.5A
Touch Panel monitor: 12V, 0.5A
Output1: Common ground
Output2: Strobe output for judgment
Output3: Output for judgment
Output4: Trigger busy
Input1: Common ground
Input2: Hold signal input for the result
Input3: Zero re-set
Input4: Start trigger (24V)
Input5: Laser on (+)
Input6: Laser on (-)
Input7: Start trigger (+5V)
Input8: Start trigger (-5V)
NPN/PNP open drain output
Withstand Voltage : Max 30V
Maximum drive current : Max 50mA
Residual Voltage Max 0.5V:
Leak current : Max 0.1mA
NPN/PNP input voltage
Max input voltage: 30V
Input current : 2.3mA
ON Voltage : Min 15V
OFF Voltage : Max 2V
Laser On (connect 5 and 6 pin: Non-voltage contact)
Internal voltage : 5V
Short-circuit current : 2mA
Input condition for OFF(Open) : Over 10kOhm of Min 4V
Input condition for ON(Short) : Max 0.5kOhm or Max 1V
Start trigger condition ( Rising edge detection for 5V system)
Input voltage range : 0֐5V
Input current : 4mA at 5Voltage
On voltage : Min 3.5V
Off voltage : Max 1V
Analog RGB

Number of Pixel

1024x768 pixel

Connector

D-sub 15pin

USB

USB 2.0 Type A for Touch panel or Mouse

Ethernet

Indicator

Environment
condition±10մInclude voltage ripple

Video Output

RS-232C
External
communications

DC24V

5JMU
.FBTVSFNFOU

Item

-BTFS
"VUPDPMMJNBUPS)JHI4QFFEBOE
)JHI3FTPMVUJPO(FOFSBM
1VSQPTF"DDFTTPSJFT

D-sub 9 pin (Cross cable)
Baud rate : 9600,19200,38400,57800, 115200
RJ-45 connecter
Transmission Method: 1000Base-T, 100Base-TX
Communication Method: TCP/IP, Socket
Data output, Control command input/output
LED(Green/Red)

Operating Temperaturerange

0֐40deg

Operation humidity range

35֐85% RH

Storage Temperature-10֐60deg

Anti-vibration

Max. 2G X/Y/Z axial direction at 10-500Hz
Appr. 900g

Note: HPU-500 is not sold standalone, Please order HPU-500 with sensor head.

5014