高精度・高スループット
２Ｄ‐３Ｄインスペクション技術マニュアル

平成２１年度九州産業技術イノベーション創出共同体形成事業
研究開発環境支援事業「高精度・高スループット２Ｄ－３Ｄインスペクション技術の開発」

＜執筆者一覧（敬称略）＞

安部 英一

産業技術総合研究所 九州センター

九州産学官連携センター

産学官連携コーディネータ
坂井 一文

産業技術総合研究所 九州センター
光計測ソリューションチーム

野中 一洋

田中 清隆

招聘研究員

産業技術総合研究所 九州センター
光計測ソリューションチーム

生産計測技術研究センター

生産計測技術研究センター

研究チーム長

産業技術総合研究所 九州センター

生産計測技術研究センター

光計測ソリューションチーム
蒲原 敏浩

産業技術総合研究所 九州センター
光計測ソリューションチーム

内野 正和

福岡県工業技術センター
強度解析チーム

重森 清史

特別研究員

機械電子研究所

専門研究員

熊本県産業技術センター

ものづくり室研究参事
久保敦

鹿児島県工業技術センター

電子部

主任研究員
尾前

宏

鹿児島県工業技術センター

電子部

主任研究員
上薗

剛

鹿児島県工業技術センター
主任研究員

生産計測技術研究センター

電子部

機械技術課

目

次

第1章

はじめに

頁

1.1 九州地域の経済規模と半導体産業の位置付け·················································· 1-1
1.2 日本らしい IC 製造のために ········································································ 1-1
1.3 半導体後工程と外観検査 ············································································· 1-2
1.4 研究開発課題と本事業の目標 ······································································· 1-4
1.5 本マニュアルの概要 ··················································································· 1-6
参考文献 ········································································································· 1-8
第2章

多焦点撮像装置による焦点深度の深い鮮明な撮像

2.1 概

要 ····································································································· 2-1

2.2 多焦点撮像装置の原理 ················································································ 2-2
2.3 多焦点撮像装置の使用方法 ·········································································· 2-6
2.3.1

2 焦点撮像装置の例 ·············································································· 2-6

2.3.2

3 焦点撮像装置の例 ·············································································· 2-7

2.4 多焦点撮像装置の画像処理 ·········································································· 2-8
2.5 実施例 ····································································································· 2-9（1）IC リード部 ·························································································· 2-9（2）ねじ ·································································································· 2-11
（3）ワイヤボンド ······················································································ 2-12（4）プリント基板 ······················································································ 2-13
（5）シリコン・ウエハー表裏面同時検査 ························································ 2-14
（6）DIP リード検査 ··················································································· 2-15
（7）リード変形瞬間の観察（高速画像の解析） ··············································· 2-15
（8）パーティクルカウンター ······································································· 2-16
2.6 仕様 ··································································································· 2-17

2.6.1 低倍率の場合 ···················································································· 2-17
2.6.2 高倍率の場合 ···················································································· 2-18
2.6.3 多焦点撮像ユニットの仕様 ·································································· 2-18
2.6.4 整備状況 ·························································································· 2-19
足 ······································································································· 2-20

補(1)

基本的な多焦点撮像装置の構成 ······························································· 2-20

(2)多焦点撮像装置の高さ測定への応用 ························································· 2-21

関連用語の解説 ······························································································ 2-22
参考文献 ······································································································· 2-23

第3章

分光位相差検出法による高さ測定

3.1 概

要 ····································································································· 3-1

3.2 高さ測定の原理 ························································································· 3-2
3.3 高さ測定装置の使用方法 ············································································· 3-6
3.3.1 高さ測定のフローチャート ···································································· 3-6
3.3.2 換算係数の求め方 ················································································ 3-7
3.3.3 対象物の測定手順 ················································································ 3-8
3.3.4 測定上の注意事項 ················································································ 3-9
3.4 実施例 ··································································································· 3-10
3.4.1 分光位相差検出法による測定例 ···························································· 3-10
（1）IC リード部 ····················································································· 3-10
（2）プリント基板の反りの測定 ································································· 3-11（3）ワイヤーボンドのワイヤー3D 形状検査 ················································· 3-12
3.4.2 全高測定例 ······················································································· 3-12（1）アセンブル後のプリント基板 ······························································ 3-12（2）全高測定用の画像例 ·········································································· 3-13
3.4.3 多焦点撮像装置との組み合わせによる高さ測定例 ···································· 3-13
3.5 仕

様 ··································································································· 3-14

足 ······································································································· 3-15

補（1）フィルター付マスクについて ································································· 3-15
（2）フィルターと照明光源 ·········································································· 3-15
関連用語の解説 ······························································································ 3-16
参考文献 ······································································································· 3-16
第4章

非合焦関数解析法による高さ測定

4.1 概

要 ····································································································· 4-1

4.2 高さ測定の原理. ························································································ 4-2
4.3 高さ測定の手順 ························································································· 4-5
4.4 実施例 ····································································································· 4-6
4.5 仕
補様 ····································································································· 4-7

足 ········································································································· 4-8

関連用語の解説 ································································································ 4-9
参考文献 ········································································································· 4-9

第5章
5.1 概

低コントラスト欠陥検出
要 ·································································································· 5-1

5.2 コントラス欠陥検出手法の原理 ···································································· 5-2
5.2.1 現状の欠陥検出方法の問題点 ································································· 5-2
5.2.2 本マニュアルで用いる検出手法の原理 ····················································· 5-3
5.2.3 スプライン関数の使用方法 ···································································· 5-4
5.2.4 欠陥候補部の補間 ················································································ 5-6
5.2.5 モチーフの定義と合成処理 ···································································· 5-6
5.2.6 補間候補の決定 ··················································································· 5-8
5.2.7 矛盾する補間候補対の決定 ···································································· 5-8
5.2.8 遺伝的アルゴリズムの適応方法 ······························································ 5-9
5.2.9 染色体の定義 ······················································································ 5-9
5.2.10

各データ点濃度の決定方法 ·································································· 5-9

5.2.11

適応度の定義 ··················································································· 5-10

5.2.12

GA の適応度の定義 ·········································································· 5-11

5.2.13

GA の進化方法と終了判定 ································································· 5-11

5.2.14判

定···························································································· 5-12

5.3 低コントラスト欠陥検出性能評価実験························································· 5-12
5.3.1 パラメータの説明 ·············································································· 5-12
5.3.2 性能評価実験条件 ·············································································· 5-13
5.3.3 性能評価実験結果 ·············································································· 5-14
5.4

IC リードフレーム検査システム ······························································· 5-16

5.4.1

IC リードフレーム検査システムの概要 ················································ 5-16

5.4.2 欠陥検出実験例 ················································································· 5-20
5.5 仕
補様 ··································································································· 5-22

足（欠陥解析） ····················································································· 5-23

関連用語の解説 ······························································································ 5-26
参考文献 ······································································································· 5-28

第6章

スペックル干渉法を利用した３Ｄ変形分布計測

6.1 概

要 ····································································································· 6-1

6.2 スペックル干渉法 ······················································································ 6-2
6.2.1 スペックル干渉法の原理[1,2] ································································· 6-2
6.2.2 スペック干渉法の数値化 ······································································· 6-4
6.3 スペックル干渉法を利用した面外変形分布計測装置の使用方法 ························· 6-7
6.4 実施例 ····································································································· 6-8
6.4.1 実施例 1 ····························································································· 6-8
IC チップを駆動させたときの面外変位分布計測結果（1） ································· 6-8
IC チップを駆動させたときの面外変位分布計測結果（2） ································· 6-9
IC チップを駆動させたときの面外変位分布計測結果（3） ······························· 6-10
6.4.2 実施例 2 ··························································································· 6-11
IC チップ（SN74LS04N）を駆動させて 5 秒後の面外変位分布計測結果（4） ···· 6-12
IC チップ（SN74LS04N）を駆動させて 10 秒後の面外変位分布計測結果（5） ·· 6-13
IC チップ（SN74LS04N）を駆動させて 5 後の面内変位（水平方向成分）分布計測結果（6） ·································································································· 6-14
IC チップ（SN74LS04N）を駆動させて 10 後の面内変位（水平方向成分）分布計測結果（7） ·································································································· 6-15
IC チップ（SN74LS04N）を駆動させて 5 後の面内変位（垂直方向成分）分布計測結果（8） ·································································································· 6-16
IC チップ（SN74LS04N）を駆動させて 10 後の面内変位（垂直方向成分）分布計測結果（9） ·································································································· 6-17
6.5 仕様 ··································································································· 6-18

関連用語の解説 ······························································································ 6-19
参考文献 ······································································································· 6-19
第7章

まとめ ································································································ 7-1

第1章

はじめに

1.1

九州地域の経済規模と半導体産業の位置付け

九州地域は、総面積（44,453km2）および総人口(1,465 万人)では全国の 12％弱（2007年）、域内総生産（GDP）では 48.1 兆円（2005 年度）で全国の 9.3％を占める。このため、九州はわが国の１割経済と概括され〔1〕、GDP では世界 16 位のオランダや 17 位のベルギ
ーに相当する大経済圏を形成している。この内、九州地域の牽引産業と位置付けられる半導体産業については、IC 生産量が全国の２割を占め（2007 年）、生産数量：8,157 百万個（全国比：21.4%）、生産金額：10,348 億円（全国比：27.5%）に達している。半導体関連(IC、半導体素子、半導体製造装置)製品（2006 年）については、出荷額：約 2 兆円（全国
シェアは約 21％）、事業所数：255 社、従業員数：46,119 人となっている。さらに、九州
における IC１個当たりの生産金額は、2000 年度をターニングポイントとして全国を上回って推移しており〔2〕、家庭用ゲーム機や自動車向け LSI など、高付加価値製品生産が寄与
している。

1.2

日本らしい IC 製造のためにIC 製造工程は、Si ウエハ上に多数の IC(チップ)を作り込む「前工程」と､ウエハを 1 個 1個のチップに切り分けた後､IC チップをパッケージに収納する以降の「後工程」の二つに大別される。この内、九州地域には、超 LSI デバイス製造工場など、中堅･中小企業を中心に
この後工程を担当する多くの地域企業が展開している。
後工程には、組立工程､仕上げ工程､選別工程､バーイン工程､および検査工程などが含まれ、各工程において入念な検査を経て製品が完成される。年々進む半導体の微細化に伴って、IC パッケージに要求される検査の許容値が微小化しており、高精度で高速な検査技術に対する要求が高まっている。さらに、従来技術では対応できない新たな外観検査のニー
ズや人間の感覚に頼った官能検査を自動化･定量化できる技術開発が必要になっている。
海外企業との熾烈な競争環境の中で、日本らしい高品質で高信頼性の付加価値の高い半導体製品の生産力を確保することは、九州地域に期待される最重要課題のひとつである。
このためには、新技術の開発･導入はもとより、その土台となる地域企業や研究機関の研究者・技術者の技術力の向上が不可欠である。これらの人材育成およびネットワーク化等を進め、地域企業の検査工程の負担軽減、高品質化、高精度化による市場競争力・国際競争力の強化が必要である。検査技術に関するこのような取り組みは、IC 産業にとどまらず、
電子機器産業や自動車産業における各種インスペクション技術への応用が可能である。

1-1

1.3

半導体後工程と外観検査

半導体後工程（組み立て工程）の製品検査においては、図 1.1 に示すように、製造過程の各ポイントで外観検査が行われ、最終検査では、図 1.2 および図 1.3 に示す様な QFP 型および BGA 型に代表される半導体パッケージの種々の外観検査が自動で行われている。そして、良品と判定された半導体パッケージはユーザの用途に応じて、様々なプリント基板に実装され、所定の機能を有する電子部材として使用されている。
：外観検査実施例＜後工程＞
ICウェハの受入

ダイシング（ウェハの切断）

ICチップ検査

チップの
マウント

ダイボン
ド検査

ワイヤボ
ンディン
グ

ワイヤボン
ディング検査

モールディ
ング（樹脂
封入）

モール
ド検査

タイバ切断
(リード不要部
の切断)

タイバ
切断検査

リードピ
ン加工

実 装

実装品検査

ICウェハ
ICチップ

ICリードフ
レーム製造

ICリード
めっき加工

リードピン
加工検査

ICリードフレーム検
査

温度電圧試験
電気試験

印 字マーキング検査

ICパッケージ
品最終検査

図 1.1 実装までの外観検査例

モールド

リード

図 1.3 BGA 型パッケージ

図 1.2 QFP 型パッケージ

このような半導体パッケージは、年々進む半導体の微細化に伴って、実装上、パッケージに要求される 2D-3D 諸元の許容値が微小化すると共に、新たな外観検査のニーズも上がってきている。そのため、様々な 2D-3D 検査項目をより高速で精度良く測定することを可能とする検査装置の高スループット化や、しみ、錆、傷、付着する異物などの認識精度の向上を一段と図り、これまでにも増して不良品の発生を低減させることが求められている。
具体的には、検査の高スループット化については、図 1.1 の IC パッケージ品最終検査に於いて、現状、ラインセンサや変位センサ等で測定されている QFP 型パッケージ（図 1.2
参照）のリードの接地面からの浮き・接地角、パッケージの反り・接地面に対する傾き、パッケージ下面・上面の接地面からの高さ等の 3D 諸元を 2D 画像から検出する方法が希求されている。これらの課題は、BGA 型パッケージ（図 1.3 参照）にも共通しており、バンプ（白い小球群）の高さ分布、パッケージの反り等の 3D 諸元を 2D 画像から検出する方法に強いニーズがある。本方法は、現状、抜き取りで目視により行われているワイアボンデ
1-2ィング検査（図 1.1 を参照） － ワイアループ高さ、ワイア間隔等の 3D 検査 － の全数自動検出に向けた課題でもある。
また、認識精度の向上については、現在、目視で行われている IC チップのマウント前の IC リードフレーム検査（図 1.1 及び図 1.4 を参照）における、しみ（メッキ漏れ、メッキ後残存薬物等）、錆、ゴミ、傷等の欠陥の自動検出の問題がある。
上記は、実装前の製品検査に関わるものであるが、プリント基板の実装（図１.1 実装品検査を参照）を行っている中小企業では、検査装置付の高価なマウンタ（実装装置）を頻繁に更新することができずに、未だに人手による外観検査や既存の検査装置を用いているところもある。従来のカメラを用いた検査はプリント基板を真上から観察・計測する方法で行われているが、この方法では部品の有無，部品の位置，部品の向き等、実装ミスの一部しか検査できない。このため、プリント基板実装時における、半導体パッ図 1.4

IC リードフレーム

ケージのリード浮き、部品浮き、クリームハンダ印刷時のパターン等の 3D 諸元を 2D 画像から検出する方法が求められている。
また、半導体パッケージは複数の熱膨張の異なる部材で構成された複合材料である。そのため、実装時や通電時（作動時）発熱による温度変化により、部材の熱膨張の違いに由来する動作不良や破損などの不具合が生じる場合がある。そこで電子部品の信頼性向上のために半導体製品の実装時や通電時（作動時）の高精度な 2D-3D 変形計測が求められている。実装時や通電時（作動時）の変形を把握することにより、動作不良や破損などの不具合が生じないパッケージの構造設計、材料選定が可能となる。

1-3

1.4

研究開発課題と本事業の目標

上記のニーズに応えるためには、画像処理のみにより質の高い位置・濃淡・高さ情報を取得する方法、並びに高精度に変形分布を計測する手段を確立しなければならない。
認識精度の向上のためには、先ず、焦点深度の深い鮮明な画像を撮ることが重要であり、次に、その画像から、しみ、傷等の欠陥や異物を認識するアルゴリズムを確立することが必要である。また、高スループット化のための 2D 画像に基づく 3D 検査においては、2D
画像から高さ情報を読みとる手法の確立が求められる。さらに、半導体の実装時や通電時（作動時）の変形計測を行うためには、温度変化によって時間的に変化する 2D-3D の膨張・収縮現象を高精度に評価する手法が必要となる。
以上の理由により、半導体後工程（組み立て工程）の中でも現場からのニーズが特に強い、IC リードフレーム検査、ワイアボンディング検査、半導体パッケージ検査、およびプ
リント基板（実装品）検査を中心に、九州産業技術イノベーション創出共同体形成事業・研究開発環境支援事業｢高精度・高スループット 2D-3D インスペクション技術の開発｣を遂行する（平成 20 年度～21 年度、経済産業省）。本支援事業は、九州イノベーション創出促
進協議会（Kyushu Innovation Creative Collaboration; KICC、平成 20 年 9 月 18 日設立）
を活動の推進母体とし、企業の技術課題への解決に資する試験・評価・分析方法を確立、それをマニュアル化し、企業への技術移転の促進および企業技術者等の技術力向上に資す
ることを目的としている。本事業への参画機関としては、これらの技術課題に関連した各地域企業への対応実績および技術ポテンシャルを有する、産総研九州センター、福岡県工業技術センター、熊本県産業技術センターおよび鹿児島県工業技術センターの 4 機関で構成し、互いに密接に連携して事業に取り組む。具体的な研究課題、目標、および分担は以下の通りである。
ⅰ．焦点深度の深い鮮明な撮像法の確立
既存撮像装置の最高 3 倍、焦点深度の深い鮮明な画像を取得可能な撮像法を確立する。
担当機関：産総研九州センター
ⅱ．2D 画像に基づく 3D 情報取得法の確立
非合焦関数解析や分光位相差検出の光計測原理を用いて、平面画像から高さ情報を取得する手法を確立する。
・半導体パッケージの場合、測定視野が約 20mm 角のとき、測定精度±10μm、測定時間 200ms／パッケージ、・プリント基板（10cm 角程）の場合、測定精度±20μm、測定時間数十秒～数分／基板、・ワイアボンディングの場合、測定精度±20μm、測定時間数百ミリ秒／パッケージ、を目標とする。本手法を用いることにより、既存外観検査系の簡略化と検査時間の短1-4縮により、検査装置の高スループット化を図る。
担当機関：産総研九州センター、鹿児島県工業技術センター
ⅲ．低コントラスト欠陥検出法の確立
上記、鮮明な画像に基づき、IC リードフレームに生じるしみ等に見られ、画像上、正常部との濃度差が尐ない欠陥検出手法を確立する。スプライン関数を用いて、現在、目視で行っている IC リードフレームのしみ検査を 95％認識率で自動検出することを目標とする。
担当機関：熊本県産業技術センター
ⅳ．高精度 2D-3D 変形計測法の確立
レーザスペックル干渉法を用いて、測定物表面に沿った面内変形（x,y 軸方向）と面
外変形（z 軸方向）を数十 nm の分解能で計測が可能な半導体パッケージの高精度動的
変形計測手法を確立する。
担当機関：福岡県工業技術センター
こうして作成した本マニュアルは、関係企業や研究機関の技術者に対し、その使用を原則オープンにする。企業や研究機関の技術者等が自ら本技術マニュアルを修得することにより、技術力向上等の人材育成に資すると共に、関連企業にマニュアルの普及を進めて行くことにより、当該企業における生産性の向上や製品不具合対応力の向上に結びつけてい
く。また、このような普及活動を通じて、新たなニーズを掘り起こし、そのソリューショ
ン提供に取り組むことで本マニュアルのさらなる進化を図っていく。

1-5

1.5

本マニュアルの概要

本マニュアルは、これまで 2 年間の本事業の取り組みの成果をまとめたものである。上記の研究開発課題および事業目標に対応して、以下の 7 章から構成される。
第 1 章「はじめに」では、半導体後工程における各種外観検査について概説し、生産現場における課題と、本事業の目的を述べた。
第 2 章「多焦点撮像装置による焦点深度の深い鮮明な撮像｣では、半導体パッケージ及びプリント基板を対象として、多焦点撮像光学系および照明系の構築、２焦点及び３焦点撮像法の開発と、そのための画像合成ソフトについて述べた。
第 3 章「分光位相差検出法による高さ測定」では、IC リード、実装基板、およびワイヤ
ボンドのワイヤの高さ計測等を例にとり、異なる色フィルター（2 個）を形成したマスクをレンズの直後に配置するだけの簡単な光学系を構築し、カラーCCD 撮像によって得られる異なる色の輝度位置の距離から対象物の高さ、または深さを測定する方法について述べた。
第 4 章「非合焦関数解析による高さ測定」では、数百万画素の撮像素子の使用により、これまで分離できなかった合焦点からのズレ（非合焦）の情報が容易に得られるようになった点に着目し、このズレに対応した 2 次元画像の広がり度合いを数値化することで微小な高さを測定する方法について述べた。
第 5 章「低コントラスト欠陥検出」では、IC リードフレームにおけるしみ等の検出を例にとり、スプライン関数を用いた欠陥特徴の解析と欠陥検出ソフトおよび装置の開発を行い、低コントラスト欠陥検出システムについて述べた。
第 6 章「スペックル干渉法を利用した 3D 変形分布計測」では、スペックル干渉法を利用した動的 3 次元変形計測法による半導体パッケージの面内および面外変形計測について述べた。
第 7 章「まとめ」では、本事業の成果を総括し、今後の展開について述べた。
なお、第 2 章から第 6 章までの各章では、概要、原理、使用方法、実施例、仕様、補足、
関連用語の解説、および参考文献の各項を記述することを共通様式とした。その中で、仕様については、それぞれの手法により“どれだけ測れるか”を簡潔に示している。
平成 20 年度には、各課題について原理実証試験装置を試作し、実験室レベルでの基本的な実施例を含んだ内容での成果報告を KICC ホームページ上で公開した。平成 21 年度には、
20 年度に開発した試作機について改善･改良を行い、現場対応に結びつけるための応用装置としてシステム化に取り組んだ。これまでにマニュアル化した画像処理ソフト等について、
検査項目ごとに生産現場に対応した実証試験を行い、半導体後工程における IC パッケージの 2D-3D インスペクション技術の確立を図ると伴に、マニュアルを完成させている。なお、本事業の最終的な総括は、平成 22 年 3 月末日であるため、ここに報告する内容の一部については、今後加筆･修正の行われることがあることを付記する。なお、本マニュアル最終版
1-6

は、web 等で公開の予定である。

1-7

参考文献
〔1〕 九州経済調査協会、九州経済サマリー
http://www.kerc.or.jp/data/download/gaikan/2009/summary.pdf〔2〕 九州半導体イノベーション協議会、半導体関連データ
http://www.siiq.jp/d01.html

1-8

第2章

多焦点撮像装置による焦点深度の深い鮮明な撮像

2.1

概

要

異なる焦点に合焦した画像を得るためには、ステージや撮像部（カメラ etc）を段階的に移動させて多数の画像を得た後、各画像にラプラシアンフィルター*などの処理を施して全ての面に合焦した画像を合成する方法が広く用いられている。この目的の装置としては、国内の多くの企業＊）から提供されている。しかし、このような装置では高価で、かつ複雑な機構が用いられており、半導体プロセスのオンライン検査装置としてはほとんど使用されていない。現状では、研究や解析用ツールとしての用途に留まっている。
本マニュアルで紹介する多焦点撮像装置は、キューブ・ビームスプリッタ*と補色フィル
ター*を組み合わせた簡易な構成によって、３つ（または２つ）の異なる焦点面に同時に合焦した像を得る簡易・ローコストの撮像装置である。
この装置は、以下の特徴をもつため、半導体プロセスの後工程への応用が期待できる。
①３つ（または２つ）の合焦位置を調整することにより、使用している対物レンズの焦点深度を擬似的に 3 倍（または２倍）に拡張できる。
②合焦する３つ（または２つ）の像は、正確に同時刻（同時取り込み）に撮像されるものであり、高速に移動する物体の撮像では、像のずれが無い。
③離散した３つ（または２つ）の物体面に合焦できる。
④駆動部がない。（焦点位置調整のための駆動部はあるが、使用中は固定）⑤電源が不要である。

注＊）たとえば、キーエンス社

2-1

2.2

多焦点撮像装置の原理

図 2.1 に多焦点撮像装置の原理を示す。図 2.1(a)は、高さの異なる３つの被写体面に焦点を合わせた場合の結像面位置を３つの色で示した。レンズの結像関係の式より、赤・緑・青の 3 つの点に同時に合焦させることはできない。このため、一般の全焦点顕微鏡は、対物レンズや撮像装置、または物体を移動させて、多数の合焦像を得る方法を用いている。
しかし、図 2.1(a)に示すように、何らかの方法によりレンズ主面から結像面までの距離を自由に変化させることができれば、高さの異なる３つの被写体面からの反射光を同一の結像面に合焦させることができる。

フィルター（Ｃ）
Ｇ
フィルター（Ｙ）

結像面

フィルター（Ｍ）
Ｂ

Ｇ

Ｒ

Ｒ
Ｇ

Ｒ

Ｂ

Ｂ

Ｒ

Ｇ

Ｂ
被写体

被写体

(b)

(a)
図 2.1 多焦点撮像装置の原理概要図2-2

この光路長の変化を実現する方法としては種々考えられるが、その中でもっとも簡易で調整が容易な方法を図 2.１(b)に示す（他の方法は、特許文献 特開 2007-295326 号を参照）。
図 2.1(b)において被写体からの反射・散乱光は対物レンズを通してキューブ・ビームスプリッタに入り、透過光と反射光に分かれて進む。反射光はマゼンダ・フィルター（M）によって緑色の光の成分のみが反射され、ビームスプリッタを通って CCD*面に結像される。マゼンダ・フィルター（M）は、その位置を左右方向に移動させることができるため、光路長を調整することができる。よって、対物レンズや CCD 面を移動させることなく、自由な位置に合焦させることができる。一方、キューブ・ビームスプリッタを透過した光は、イエロ・
フィルター（Y）によって青色成分のみが反射され、緑色および赤色成分はそれぞれ透過する。イエロ・フィルター（Y）はビームスプリッタによって固定されているため、青色成分の光はレンズと CCD の位置によって固定された面に合焦する。イエロ・フィルター（Y）
を透過した光はシアン・フィルター（C）によって赤色成分のみ反射される。シアン・フィルター（C）はマゼンダ・フィルター（M）と同様に移動可能なため、赤色成分の光路長は任意に調整可能である。反射された青色成分と赤色成分はビームスプリッタで反射されてCCD 面に結像される。このように１つの固定フィルターと２つの稼動調整の可能なフィル
ターを用いて、高さの異なる３つの被写体面の任意の位置に合焦した画像を同時に撮像できる。
多焦点撮像装置では、３つの色成分について個々にその光路長を調整することによって
CCD 面に同時に結像させることをひとつの特徴としている。このため撮像装置には、カラーCCD カメラを使う必要がある。CCD の RGB の画素を用いて、それぞれに合焦した画像を形成させている。よって、カラーCCD で得られる画像から、R、G、B 成分の画像を分離し、画像処理（アフィン変換*：倍率や平行移動・回転）した後に、画面合成する必要がある。この処理については「2.4 多焦点撮像装置の画像処理」で述べる。
また、3 色について色毎に別々の画面を構成しているため、フィルターの分光特性が CCDの分光特性とほぼ一致している必要がある。図 2.2 にフィルターの分光特性を、図 2.3 にCCD の分光感度特性の例をそれぞれ示す。使用する CCD の分光感度特性に一致するフィ
ルターの選択が、本装置のひとつのポイントと言える。
フィルターの分光透過特性（図 2.2）および CCD の分光感度特性（図 2.3）からわかる
ように、それぞれの赤色成分、緑色成分、および青色成分は互いに一部の波長域に重なりを持っている。例えば、図 2.3 に示すように CCD カメラのＲ、Ｇ、Ｂの各画素子のもつ分光特性については、緑色成分は赤色成分および青色成分と一部の波長域に重なりを持つために、この緑色成分の一部の重なった波長域の光についてはＲおよびＢの各画素子で受光されることを示している。本装置では色によって焦点位置を区別しているため、色が混ざり合うと、焦点がボケた画像が得られることになる。この効果は一般の CCD やフィルター
では除くことができない。これを回避するためには、照明光に工夫が必要である。
現在、照明光として低消費エネルギーの LED が多く使用されているため、ここでは白色
2-3

LED を使用した場合について述べる。白色 LED は３種類が販売されている。紫外 LED を発光させて蛍光膜を照射し、その発光によって白色光を得る方法、同様に青色 LED を用いて、白色光を得る方法がある。これらは可視域全体に広がる連続光であるため、上記問題を解決できない。一方、R、G、B の３つの LED を１つのパッケージに収めた白色光源が販売されている（図 2.5，2.6）。これは、それぞれの LED の発光特性を重ね合わせた発光
分布をもっており、図 2.6 のように RGB ごとに狭帯域の照射光を得ることができる。このLED を用いることによって、上記のフィルターによる重なりを低減できる。その他、連続スペクトルに、狭帯域フィルターを施す方法や、高価になるがフィルターを蛍光顕微鏡などで使われている狭帯域フィルターに変更するなどの手法が考えられる。
100

ｼｱﾝ
透
過 50
率

580nm ～

0
300

550

波長（nm)

800

100

ﾏｾﾞﾝﾀﾞ
透
過 50
率500～580nm
0
300

550

波長（nm)

800

100

ｲｴﾛｰ
透
過 50
率

～510nm0300

550

波長（nm)

図 2.2 フィルターの分光透過特性
2-4

800

610nm

540nm

460nm

100

Red

80

感
度

Green
Blue

60

%
40

20

0
400

500

600

700

波長（nm)

図 2.3 ＣＣＤの分光感度特性の例

Ｇ Ｂ

Ｇ Ｂ

Ｒ Ｇ

Ｒ Ｇ Ｒ Ｇ

Ｇ Ｂ

Ｇ Ｂ

Ｒ Ｇ

Ｒ Ｇ Ｒ Ｇ

Ｇ Ｂ
Ｇ Ｂ

図 2.4 単板式カラーＣＣＤ ベイヤ型素子配列の例

460nm 530nm
白

色

630nm

光

相
対
強
度
赤色
LED

緑色
LED

青色
LED

波長（ｎ ｍ）

図 2.5 赤・緑・青色 LED による白色 LED2-5

図 2.6 白色 LED による発光スペクトル2.3

多焦点撮像装置の使用方法

2.3.1

2 焦点撮像装置の例

図 2.7 は高さの異なる 2 ヶ所に同時に焦点が合った画像を得る装置の構成図である。
〔操作手順〕① レンズの光軸に合わせ、被写体を適切な位置に置く。
② 被写体のＰ0 の位置に赤成分の光で焦点が合うように装置と被写体の距離を調節する。
この時、モニターで赤成分の分解画像を見ながら行う③ ねじを回しマゼンタ・フィルター(M)の位置を調節してＰ0 の位置に緑成分の光で焦点が合うようにし、この位置を基準位置とする。この時、モニターで緑成分の分解画像を見ながら行う。

④ 続いてマゼンタ・フィルター(M)を被写体から離れる方に移動させＰ1 に焦点を合わせる。
（逆方向に移動させるとＰ2 に焦点が合う）この時、モニターで緑成分の分解画像を見ながら行う。

⑤ この状態で撮像を行うとＰ0 の位置に焦点が合った赤成分画像とＰ1 またはＰ2 に焦点が合った緑成分画像が同時に得られる。

モ ニ タ ーね じ
フ ィ ル タM）ー （Ｇ

メ モ リ ーＧＲ

Ｂ

撮 像 制 御 部

ビ ー ム ス プ リ ッ タ

撮 像 素 子フ ィ ル タC）
ー （
Ｇ Ｂ Ｒ
Ｐ１
Ｐ０
Ｐ2

被 写 体

図 2.7 ２焦点撮像装置の概要と合焦像の取得法
2-6

2.3.2

3 焦点撮像装置の例

図 2.8 は高さの異なる 3 ヶ所に同時に焦点が合った画像を得る装置の構成図である。
〔操作手順〕① レンズの光軸に合わせ、被写体を適切な位置に置く。
② 被写体のＰ0 の位置に赤成分の光で焦点が合うように装置と被写体の距離を調節する。
この時、モニターで赤成分の分解画像を見ながら行う。

③ ねじＡを回しマゼンタ・フィルター(M)の位置を調節してＰ0 の位置に緑成分の光で焦点が合うようにし、この位置を基準位置とする。この時、モニターで緑成分の分解画像を見ながら行う。

④ 同様にねじＢを回しイエロ・フィルター(Y)の位置を調節してＰ0 の位置に青成分の光で焦点が合うようにし、この位置を基準位置とする。この時、モニターで青成分の分解
画像を見ながら行う。

⑤ ねじＡでマゼンタ・フィルター(M)の位置を調整し緑成分の光でＰ1 に焦点を合わせる。
⑥ 同様にねじＢでイエロ・フィルター(Y)の位置を調整し青成分の光でＰ2 に焦点を合わせる。

⑦ この状態で撮像を行うとＰ0、Ｐ1、Ｐ2 の 3 ヶ所に焦点が合った画像が同時に得られる。
ねじB
モニ タ ー
モニター
フィルター(Y)メ モ リ ー
メモリーフィルター(C)

ねじA

B

B
G

R

撮 像 制 御 部
撮像制御部

R
ﾋﾞｰﾑｽﾌﾟﾘｯﾀ

撮 像 素 子
撮像素子

フィルター(M)対 物 レ ン ズ
対物レンズ

ビームスプリッタ
Ｂ

Ｇ

Ｒ

Ｐ2
Ｐ1

被写体

Ｐ0

図 2.8 ３焦点撮像装置の概要と合焦像の取得法

2-7

2.4

多焦点撮像装置の画像処理

多焦点撮像装置で得られる画像は、全焦点カメラと異なり 2 点または 3 点に合焦しているため、全焦点カメラとその合成法は多尐異なるが、基本的な画像合成手順は同じである。
図 2.9 に多焦点画像を得るフローを示す。カメラより画像を取得した後、各色画像に分離
し 3 画像のうち 2 画像（例えば、青、緑画像）は、微分フィルターなどを通し、合焦領域を判断する。それぞれの合焦領域を加算し、残りの領域をもう一枚の画像（例えば赤画像）
で埋めることによって合成画像が得られる。（多焦点撮像装置は、非合焦領域が含まれているため、全焦点カメラのように合焦領域のみを加え合わせた場合に、画像が抜ける領域が生ずる）また、各色画像は画像倍率が異なるため、必要に応じてアフィン変換を行う。
S T A R T

画 像 取 得多 焦 点 画 像

色 画 像 に 分 割

青
画 像
青 画 像
赤 画 像

緑 画 像

青 画 像

ﾟ ﾗ理
ｼ ｱ ﾝ ・ ﾌ ｨ ﾙ ﾀ 処 理
ﾗ ﾌ ﾟ ﾗ ｼ ｱ ﾝ ・ ﾌ ｨ ﾙﾗ ﾀﾌ 処

緑 画 像

青 画 像

合焦
焦 領領域域
青 ２のの値合
ﾓ ﾙ ﾌ ｫ ﾛ ｼ ﾞ ｰ 変 換 ﾓと ﾙ２ ﾌ値ｫ ﾛ ｼ ﾞ ｰ 変 換 青と合 焦 領 域 の 判 別

合 焦 領 域 の 判 別

緑 の 合 焦 領 域

青 の 合 焦 領 域

青
青のの 合合焦焦領領域域
ア フ ィ ン 変 換
赤 画
画 像
赤
像
合 焦 画 像EN

D

図 2.9 画像処理のフローチャート及び画像
2-8

2.5

実施例

多焦点撮像装置によって得られた画像例を示す。
（1）IC リード部
図 2.10 と図 2.11 は多焦点撮像装置で同時に取得した赤画像と青画像とを画像合成処理
により合成した画像である。また、図 2.12 は赤画像と緑画像の光軸を青画像からずらし、
1 箇所の IC リード部を一つの画像上で取得し、画像を合成したものである。

リード部の肩口に合焦した赤画像

リード部の先端に合焦した青画像

リード部の先端に合焦した青画像

リード部の肩口に合焦した赤画像

合成処理

合成処理

合成画像

合成画像

図 2.10 IC リードの多焦点像（1）

図 2.11

2-9

IC リードの多焦点像（2）リードの先端（青）・中間付近（緑）・グレー画像に変換した画像

肩口（赤）に合焦した画像

グレー画像をフィルター処理した画像

フィルター処理した画像を二値化二値化した画像から合焦領域を判別

合成画像

図 2.12

IC リードの多焦点像（3）2-10

（2）ねじ図 2.13 は、ねじの先端と頭部に合焦させた画像である。合焦位置の差は約 30mm である。
但し、画像では見えないが、先端と頭部の間は焦点が合っていない。

撮像方向

ねじの先

30mm

ねじの頭
ねじの先に合焦した像

合
成
処
理

ねじの頭に合焦した像

合成画像

図 2.13 ねじの先端と頭部の多焦点像

2-11

（3）ワイヤボンド
IC のワイヤボンド*部の 2 焦点像。
（ワイヤは一部、切れて浮き上がっている）右下の画
像は、中間のワイヤを跳び越して、電極と上側のワイヤに焦点を合わせた 2 焦点像である。
尚、これらの画像は多焦点装置で得られる未処理の画像であり、画像合成を行っていない。
高倍率の場合、未処理でも短焦点画像より良好な画像が得られる。
【単焦点画像】【２焦点画像】

ワ イ ヤ に 焦 点 が 合 っ た も の

ワ イ ヤ と 電 極 に 焦 点 が 合 っ た も の

電 極 接 合 部 に 焦 点 が あ っ た も の

電 極 と 接 合 部 に 焦 点 が 合 っ た も の

ワ イ ヤ に 焦 点 が 合 っ た も の

ワ イ ヤ と 電 極 に 焦 点 が 合 っ た も の上 の ワ イ ヤ に 焦 点 が 合 っ た も の

異 な 2る
本 の ワ イ ヤ に 焦 点 が 合 っ た も の上 の ワ イ ヤ と 電 極 に 焦 点 が 合 っ た も の

図 2.14 IC のワイヤボンド部の多焦点像（未合成処理画像）
2-12

（4）プリント基板
図 2.15 及び図 2.16 はプリント基板の画像の多焦点像で、それぞれ 6 枚の画像を合成し
ている。

ピント：上

多焦点像

合成

ピント：下

《レンズ等の構成》
CCD

レンズ

多焦点撮像装置

プリント基板

図2.15 プリント基板の多焦点像 (1)

ピント：上

多焦点像

合成
ピント：下

《レンズ等の構成》
CCD

多焦点撮像装置

レンズ

プリント基板

図2.16 プリント基板の多焦点像 (2)2-13

（5）シリコン・ウエハー表裏面同時検査
多焦点撮像装置のフィルターを赤外カットフィルターに変更することで、赤外と可視の２焦点撮像装置が可能となる。この場合、CCD カメラには、白黒カメラ（赤外カット
フィルターを取り除く）を用い、画像はフィルターの煽り調整を用いて、画面を２分割し、可視画像と赤外画像を同時に表示する。構成例を図 2.17 に、パターン付き上ウエハの表裏同時観察例を図 2.18「に示す。このように可視光だけでなく赤外線や紫外線などの画像への応用も容易である。

図 2.17 赤外と可視画像を取得する多焦点撮像装置の構成例

図 2.18 赤外＆可視によるパターン付きウエハの表裏同時観察例

2-14

（6）DIP リード検査
DIP タイプの IC は、リード列の間隔が広く、通常の光学系では、同時にすべてのリードにピントを合わせることはできない。また、その途中にピントを合わせる必要はなく、離散点に合焦すればよく、多焦点撮像装置の最も適した使用法でもある。図 2.19 には、
通常の光学系による撮像画像例（上）を、多焦点撮像装置を用いた画像（中、下）を示す。

従来の撮像系による画像

多焦点画像

図 2.19 赤外＆可視によるパターン付きウエハの表裏同時観察例（7）リード変形瞬間の観察（高速画像の解析）
一般に、全焦点カメラは、焦点位置を移動させて複数枚の画像を重ねて全てに焦点が合った画像を合成するため、画像間にはタイムラグが生ずる。よって、対象物が移動するする場合は、画像のズレを生ずる。多焦点撮像装置は、各色画像を同時に取得するため、画像間のタイムラグがない。よって、高速の現象でも画像ズレがない鮮明な像が得られる。図 2.20 は装置との干渉によってリードが変形する瞬間を捕らえた画像である（画像は 2 焦点画像の合成像である）。1/250 秒の形状が鮮明に捕らえられる。

2-15

図 2.20 250fps で捕らえられたリード変形の瞬間（多焦点画像合成後の画像）（8）パーティクルカウンター
溶液中のパーティクルは暗視野光学系で観察でき、その大きさは光散乱の強度から推測できる。しかし、粒子の拡散の速度は大きいため、全焦点カメラで撮像した場合、線状の画像が得られ個々のパーティクルを分解することは難しい。前述のように多焦点撮像装置では、高速撮影が可能なため複数の箇所のパーティクルを同時に計測できる。図2.21 は 532nm と 475nm のレーザーで水槽の異なる場所（深さ方位に約 10mm 離れた箇所）を照射し、暗視野系で観察したパーティクル像（高速カメラで撮影：60fps）である。
離れた箇所に同時に合焦し、個々のパーティクルが分離して観察されることが分かる。

475nm

532nm

図 2.21 60fps で捕らえた水中のパーティクルの多焦点像

2-16

2.6

仕

様

製品化された多焦点撮像装置（ＭＦＵ）の写真を図 2.22 に示す。ここで述べる仕様は MODEL MFU2 に関するものである。
図 2.22 MODEL MFU２
2.6.1 低倍率の場合
焦点深度（DOF）を測定する光学系を図 2.23 に、各色画像で得られる DOF チャート像を図 2.24 に示す。赤画像 1,2,3 は、MFU2 ユニットのフィルター位置（調整ネジ）を移動して得られた画像である。また、青、赤画像の焦点深度*とフィルター位置の移動よる合焦位置の変化を図 2.25 に示す。
青画像

赤画像１

合焦中心55.0mm

赤画像２

合焦中心53.5mm

赤画像３

合焦中心48.5mm

図 2.23 測定装置（低倍率）

合焦中心44.3mm

図 2.24 各位置の DOF チャート像

赤の合焦点範囲 4.7mm

9.1mm

青の合焦点範囲 4.7mm

図 2.25 フィルター調製量と合焦範囲
2-17

2.6.2 高倍率の場合
顕微鏡用対物レンズを用いた場合の DOF 測定光学系を図 2.26 に、測定結果を図 2.27 に示す。

赤の合焦点範囲
赤の合焦点範囲

青の合焦点範囲
青の合焦点範囲

図 2.26 測定装置（高倍率の場合）

図 2.27 フィルター調整量と合焦範囲

2.6.3 多焦点撮像ユニットの仕様【多焦点撮像ユニット（MODEL MFU2）の焦点深度仕様】
視野（mm）

12.7

3.4

実倍率

0.37

1.4

焦点深度(mm)50.5

焦点深度の最大移動量(mm)

9.1

2.1

※ 落射照明は被写体（試料）と MFU の間に入れる

2-18

2.6.4 整備状況
本事業の多焦点撮像装置に関する研究開発環境は、鹿児島県工業技術センターに整備している。多焦点撮像装置(MFU3)を用いた構成例と構成部品を図 2.28 に、試作したプログ
ラムの画面構成例を図 2.29 に示す。
区分

品名

メーカー(型式)

カメラ

USB2.0 カメラ

アルゴ(DFK31AU03)

80 万画素カラー
CCD 、 解 像 度
1024×768

レンズ

対物レンズ

中央精機(OB-3)

倍 率 :3 倍 (2.6.2
高倍率で使用)照明

メタルハライド
ファイバー光源エドモンド・オプテックス・ジャパン(MH-100)

100W

ステージ

Z軸
ステージ

シグマ光機
(SGSP60-10ZF)

移動量:10mm(1μm/パルス)

XY 軸
ステージ

シグマ光機
(TSDM40-15X)

移動量:15mm(0.5μm/パルス)

回転
ステージ

シグマ光機
(KSP-606MR)

移動量:粗 360°微±5°USB2.0カメラ
多焦点撮像装置
対物レンズ
対象
回転ステージ
Z軸ステージXY軸ステージ

メタルハライド
ファイバー光源仕様等

図 2.28 多焦点撮像装置(MFU3)を用いた構成例と構成部品

図 2.29 試作したプログラムの画面構成例

2-19

補
(1)

足
基本的な多焦点撮像装置の構成

図 2.A1 はビームスプリッタ、ミラー、原色フィルター、光路長変更用素子*によって構成した多焦点撮像装置の例である。この場合、光路長を調整するために屈折率が 1 より大きい透明材料（例えばガラス）を所望の厚さにして各色光のパス上に挿入している。このため、光路長の変更はできないが、原色フィルターを用いて焦点位置を固定した用途に、簡易に構成できる。１つのパッケージに固定できるため、振動が多い場所での使用には有効である。

図 2.A1 光路長変更用素子を用いた多焦点撮像装置の構成例

2-20

(2)

多焦点撮像装置の高さ測定への応用

図 2.A2 は被写体の高さを測定する機能を備えた例である。

パルスモータでフィルターを移動させることにより光路長を調整する機能を備えていて、パルスモータの回転量から高さを測定するものである。尚、合焦位置を判断する画像処理などが必要である。

ｴﾝｺｰﾀﾞ
モニター

ﾊﾟﾙｽﾓｰﾀフィルター(M)
メモリー

Ｇ

Ｇ

Ｒ

撮像制御部

Ｒ
撮像素子
フィルター(C)
Ｇ
Ｐ1
Ｐ0

Ｒ

対物レンズ

被写体

図 2.A2 多焦点撮像装置を用いた高さ測定法の概要図

2-21

関連用語の解説
ラプラシアンフィルター
鮮鋭化フィルターの一種で原画像の二次微分を求め、原画像からこの結果を差し引くことにより、エッジの部分の濃度の変化を強調し鮮明な画像を得ることができる。
ビームスプリッタ
一つのビームを二つ、もしくはそれ以上の分離したビームに分割する光学器具。簡単なビームスプリッタは、異なった方向にビームの一部が変えられるようにある角度でビームに挿入した極めて薄いガラス部材から成っていて、更に複雑な形では、二つの直角プリズ
ムをそれらの底面同士接合しているものがある。
補色フィルター
可視光のうち特定の波長域を通過させ特定の波長域を反射するフィルター。原色フィル
ターと補色フィルターがある。原色フィルターは光の三原色と呼ばれる赤（R）、緑（G）、青（B）で構成されたものであり、補色フィルターは三原色の補色にあたるイエロー、マゼ
ンタ、シアンを使っている。一般的には、原色フィルターの方が発色は良く再現性に優れるが、補色フィルターは感度が高く落ち着いた色合いになりやすい。
ＣＣＤ
Charge Coupled Device（電荷結合素子）の略。光を電荷に変換する半導体の一種で、デジタルカメラの撮像素子に多く用いられる。RGB 画素の配置はある行をＲＧ、ＲＧ、次の行をＧＢ、ＧＢとしたベイヤ型素子配列をしたものが多く使用されている。
撮像素子
レンズから入ってきた光を電気信号に変換する電子部品のことである。撮像素子の代表的なものとしては、CCD イメージセンサーや CMOS イメージセンサーなどがある。
アフィン変換
幾何学的な線型変換と平行移動の組み合わせによる図形や形状の移動、変形方式。4×4の行列演算で表現できる移動、回転、左右反転、拡大、縮小、シアーの座標変換。アフィン変換は元の図形で直線上に並ぶ点は変換後も直線上に並び、平行線は変換後も平行線であるなど、幾何学的性質が保たれる変換方式。
ワイヤボンド半導体チップの電極端子とリードフレームのインナーリードとの間を直径 25μ程度の金線で接続すること。
焦点深度（ＤＯＦ/Depth of Focus）
被写体の点に焦点を合わせた時、ぼけを感じない範囲のこと。
光路長変更用光学素子
屈折率の大きい透明な板状部材からなり、厚さをｄ、屈折率を n とすれば光路長は nｄとなり d(n-1)だけ長くなる。光路長変更手段のひとつ。
2-22

参考文献
〔1〕坂井一文：“多焦点撮像装置”,産業技術総合研究所 特開 2007-295326〔2〕坂井一文：“多焦点撮像装置”,産業技術総合研究所 特開 2008-059121〔3〕坂井一文：“多焦点画像を撮像する方法及び多焦点撮像装置”,
産業技術総合研究所

特開 2008-059120

2-23

第3章

分光位相差検出法による高さ測定

3.1

概

要

半導体や電子部品の外観検査では、リードの有無、リードの曲がり、モールドの欠け、マークの有無などの 2 次元検査の他に、リードの浮きやコプラナリティなどの 3 次元検査が行われている。前者は、画像処理の発達によって多くの課題が解決されてきたが、3 次元検査では解決されていない課題が多く、特にインポケットでの高さ測定は、十分な性能が得られていない状況である。また、情報端末の薄型化によってプリント基板の反り、基板から部品上面までの高さ（全高測定）などの計測が要求されている。
IC パッケージのリードの接地面からの浮き、上面・下面の高さ、バンプ*の高さ、ワイヤ
ループ*の高さなどの測定にはラインセンサ*、レーザ変位計*を用いた三角測量法、共焦点法、白色干渉法*などがある。このうち共焦点法や干渉法を用いた方法は高精度であるが、製造工程では実用的ではない。また、三角測量法（レーザー変位計など）はケースがレー
ザー・ビームの障害になるためインポケットでの検査はできない｡
本マニュアルで紹介する高さ測定装置は、位相差検出法と色情報とを組み合わせた方法
で、カラーCCD 撮像によって得られる異なる色の輝度位置の距離から対象物の高さ、または深さを測定するものである。
この装置は次のような特長を持ち、半導体プロセスの後工程への応用が期待できる。
①

一回の撮像で対象物の高さを測定できる。

②

検査に要する時間を短くしスループットを向上できる。

③

インポケット検査が可能である。

④

検査装置の構成を簡単なものにできる。

3-1

3.2

高さ測定の原理

図 3.1 に分光位相差検出法による高さ測定の原理を示す。概略的にはナイフエッジ法*と三角測量法を組み合わせたものであり、位相差検出法と同じ光学的手段を利用して高さの測定を行うものである。
図 3.1（ａ）はレンズの直後にフィルター付マスクを配置し、物体平面の像を基準像平面上に結像したものである。（レンズの焦点距離をｆ、レンズから物体平面までの距離をａ、レンズから基準像平面までの距離をｂとする）このような場合、対象物からの光はそれぞれのフィルターで透過され、基準像平面上に結像することになる。
一方、図 3.1（ｂ）はそれぞれフィルターを透過した光が基準像平面より⊿ｂだけ離れた位置で合焦した図である。この場合、基準像平面（CCD 位置）ではそれぞれのフィルターを透過した光は一つの画像にはならず、距離ｄだけ離れた色画像が得られる。

d
⊿b

ＣＣＤ
基準像平面

b

Ｄ

b

青色透過フィルター

Ｄ

赤色透過フィルター

マスク

a

a
⊿a

物体平面

(a)

(b)

図 3.1 高さ測定の原理概要図3-2

図 3.2 はフィルター付マスクの平面図である。マスク自体は遮光性の材料であり、図のよ
うに光軸に対し一対の窓を設けフィルターが装着されている。ここでは赤色透過フィルターと青色透過フィルターとし、中心間距離をＤとする。

D
マスク
赤色透過フィルター

青色透過フィルター

図 3.2 フィルター付マスクの平面図

図 3.1（ａ）において、ａ、ｂ、ｆは結像関係にあるから

1 1 1
 
a b
f

（3.1）

となる。
同様に、図 3.1（ｂ）より

1
1
1


a  a b  b f

（3.2）

が成り立つ。変形して

1
a 

a 1 

a 




1
b 

b1 

b 




1
f

（3.3）

が得られる。⊿ａと⊿ｂはａ、ｂに比較して非常に小さいとして近似すると

1 a 1 b 1
 
a a2 b b2f（3.4）

が得られる。よって式（3.4）と式（3.1）から

b 

b2
a
a2

（3.5）

の関係式が得られる。
3-3

図 3.1（ｂ）において、光が基準像平面を通過する時の距離（CCD での距離）をｄとす
ると、距離ｄと幅 D をそれぞれ底辺とする三角形の相似関係と式（3.5）から
2b  a・D
b・D  a 
ba・D
b
d



a・D
2
2
2
b  b
a

b

a
a
b
b    a
a

（3.6）

が得られる。
ここで式（3.6）により、距離ｄは微小な高さ⊿ａに比例することが分かる。また、ｄと⊿ａとの比例関係を示す換算係数はａ、ｂ、Ｄの値から求めることができるが、本マニュ
アルでは測定によって換算係数を求めている。その求める方法は「3.3

高さ測定装置の使用方法」で詳しく述べる。
図 3.1（ｂ）において基準像平面上に距離ｄだけずれた色の像が形成される。このような結像位置から、はずれた像は若干のボケが生じるが輝度確認ができればよく、およそ焦点深度の 3 倍程度までは測定できる。対象物としては、面が基準となる平面を有し、その中に凸部、凹部が存在するものであり、リード、ボール、バンプ等が該当する。

3-4

図 3.3 は異なる色で分離した像のずれを表したものである。図 3.3（ａ）は電子部品表面の一部で 2～3ｍｍ程度の範囲を示し、表面に凸部が配列されたものである。この資料で基準平面に焦点を合わせ、本装置で撮像したものが図 3.3（ｂ）である。高さをもつ凸部が、それぞれのフィルターを透過した光によって図 3.3（ｂ）のようにｄだけずれた像になり、式（3.6）によりこのｄは凸部の高さに比例したものである。

ｄ
凸部

青画像
赤画像

（ａ）

（ｂ）

図 3.3 電子部品表面の例および本装置で撮像された例

3-5

3.3

高さ測定装置の使用方法

3.3.1

高さ測定のフローチャート

図 3.4 に分光位相差検出法による高さ測定のフローを示す。測定の前処理として検量線を測定し、換算係数を求める必要がある。換算係数は式（3.6）に示すように光学系（対物レンズ、フィルター、倍率など）によって決定されるため、光学系の変更を行った場合のみ必要となる。（換算係数は色画像間距離を高さに変換するために必要な係数である。詳細は 3.3.2 を参照のこと）画像取得後、フィルターに応じて色画像（図 3.4 では、青と赤）に分離し、測定する箇所の画像間距離（画素）を求め、換算係数を掛けて高さを算出する。

S T A R T
START

検 量 線 測 定
検量線測定
換算係数を求める換
算 係 数 を 求 め る

取得画像

画 像 取 得画像取得

赤 青
赤・青画像に分離
・ 画 に
像分離

赤画像

青画像
測定する箇所の
画像間距離（画素）を計算（方法：重心、フーリエ変換etc）

赤画像

青画 像
青画像

高さ＝
画像間距離×換算係数

E N D
END

図 3.4 高さ測定のフローチャート
3-6

3.3.2換算係数の求め方

図 3.5 は分光位相差検出法を用いた装置の構成例である。最初に、異なる色で分離した
画像間距離ｄと凸部の高さとの換算係数を求めるために下記操作を行う。
①

載置ステージ面に識別部（格子模様）＊）1 を付ける。この時、対象物は載せない。

②

ロックを解除し、ノブの調整により撮像装置を上下に移動させ、載置ステージ面に焦点を合わせる。終了後、ノブを再ロックする。

③

この状態を保持したまま、移動調節手段＊）2 で載置ステージを上に移動させて載置ステージ面を撮像する。

④

得られた画像を赤・青画像に分離し、格子の各点の距離ｄを画像の座標位置から求める。

⑤

載置ステージの移動距離を記録する。

⑥④、⑤を繰り返し実行し、画像間距離ｄと載置ステージの距離との関係データを求める。

⑦

取得したデータにより各点の検量線を求め換算係数を算出する。
（図 3.6 にパルスステージの移動量と計測値（画面中央）の関係を示す）

モニター

腕部
ロ ッ ク 用 ノ ブ
ロック用ノブ
撮像部

解析処理部
ﾌｨﾙﾀｰ付ﾏｽｸ
撮像用レンズ支柱
レンズ鏡胴
載置ｽﾃｰｼﾞｽﾃｰｼﾞ駆動手段（上下に微細駆動する）基台

図 3.5 分光位相差検出法による高さ測定装置の構成図

3-7

7
6
5

計
測
値 4
（
任
意 3
）

検量線

2
1
0
０

100

200
300
パルスステージ移動量（μ m）

400

500

図 3.6 ステージ移動量とｄの計測値＊）1

識別部（格子模様）ここで用いる識別部は異なる色のフィルターで分離した像が識別し易い色（例えば白色）のものとする。

＊）2

移動調節手段載置ステージを上下に微細移動させる装置。ナノレベル（数十 nm～百nm）の測定ではピエゾ効果 ＊を利用したピエゾステージを用いる。ミクロンレベル（数十μm～数百μm）の測定ではパルスモータによるパルスステージを用いる。

3.3.3 対象物の測定手順①

図 3.5.において、対象物を載置ステージ上に置く。

②

ノブで撮像装置を調節し、対象物の基準表面に焦点を合わせる。

③

撮像装置により対象物を撮像する。

④

測定箇所の色画像間隔を画像解析により求める。

⑤

画像間距離と換算係数とから測定箇所の高さを求める。

3-8

3.3.4 測定上の注意事項
光学系はそれぞれ特有の収差が含まれている。分光位相差検出法を使う場合は、特に歪曲収差が大きく影響する。正方格子を撮像した場合、図 3.7 のような画像が得られる。
図 3.7（ａ）を糸巻型歪曲、図 3.7（ｂ）を樽型歪曲という。

（ａ）糸巻型歪曲（ｂ）樽型歪曲

図 3.7 正方格子を撮像した場合の歪曲収差

このような光学系で分光位相差検出法の測定を行った場合、図 3.8 のように曲線上に測定点が並ぶ。このため歪曲を予め測定し補正する必要がある。

補正前のデータ

高さ
（μ m）

Pin番号
図 3.8 歪曲収差の影響を受けた測定値
3-9

3.4

実施例

3.4.1 分光位相差検出法による測定例（1）IC リード部
図 3.9 は赤と緑のフィルターを用いた撮像画像と先端の高さを測定したデータである。

図 3.9

IC リード部の分光位相差検出法の画像と高さ測定データ

3-10

（2）プリント基板の反りの測定
プリント基板には、部品実装後に切り離しができるようスリットやＶカットが施されているが、部品実装時の加熱等により変形が生じている。画像計測を行う際は、この変形の影響が尐なくなるよう治具で固定しているが、変形を完全に抑えることができない。そこで、前述の高さ測定装置を用いて、この変形の一部を観察した。
基板上のシルク印刷（白色）を識別部とし、載置ステージの移動調節手段には、パルス
ステージ（シグマ光機製 TSDM）を用いた例を図 3.10 に示す。

レンズ
スリット

照明

シルク印刷

Ｖカット

識別部

固定治具

移動ステージ

分
離
し
た
像
の
間
隔
（
任
意
）

26
24
22
20

換
算 400
係
数
か 300
ら
算 200
出
し 100
た
高
0
さ
（
任
意
）

y = 0.0106x + 14.387
2

R = 0.9995

18
16
14
0

200

400

600

800

1000

0

5
10
15
プリント基板の位置(mm)ステージ移動量（パルス数）

図 3.10 分光位相差検出法による基板の反りの測定例

3-11

20

（3）ワイヤーボンドのワイヤー3D 形状検査
前述のように本方法は、2D 画像に 3D 情報を組み込むことが出来るため、ワイヤーの 3D
形状が容易に得られる。図 3.11 に、ワイヤーの計測例を示す。
・x,y 座標：色画像の対応する各点の中点・ｚ座標：色画像の対応する点の距離

図 3.11 ワイヤーの分光位相差検出法による画像（左）と、白枠内のワイヤー3D 形状（右）3.4.2 全高測定例（1）アセンブル後のプリント基板
図 3.12 は部品実装後のプリント基板上部品の高さを測定した例である。シルク印刷に合焦（基準高さ）し、各部分の赤と青との距離を求め、高さを算出した。
Ｄ

Ｂ

Ｃ

E
A

図 3.12

測定箇所

高さ（μm）IC パッケージ(A)1100

ﾘｰﾄﾞ先端(B)226

ﾘｰﾄﾞ根元（C)677

半田ｴｯｼﾞ(D)141

CHIP 部品(E)809

プリント基板上部品の高さ測定

3-12

（2）全高測定用の画像例
図 3.13 にプリント基板上の部品の高さ測定用画像を示す。各位置の色画像の差から部品の各部の高さが計算できる。図 3.13(b)と(c)は同じ箇所で合焦位置を変えて撮像した画像である。このように基準面は任意に設定することができる。
（ａ）基板に合焦した画像（可変抵抗）（ｂ）部品上面に合焦した画像

図 3.13（ｃ）基板に合焦した画像

プリント基板上部品の高さ測定用画像例

3.4.3 多焦点撮像装置との組み合わせによる高さ測定例
分光位相差検出法は 2D 画像に 3D 情報を色画像として組み込む手法であるが、合焦位置から外すことを前提としているため 2D 検査の画像は得られない。しかし、多焦点撮像装置と併用した擬似的なステレオ法を構成することによって、2D と 3D 検査が可能となる。図3.14 にその構成と、エンボス内の IC のリードのコプラナリティーの測定例を示す。

図 3.14 多焦点と分光位相差検出を組合わせた擬似ステレオ法の構成（左）と、エンボス内のリードの高さ測定例（右）（50 回測定）

3-13

3.5

仕

様

測定する高さは、式（3.6）を変形すると

d

a
b
D
a・D  M2aa

 a 

（ M 

b
）
a

（3.7）

d・a
M・D

（3.8）

で与えられる。よって測定可能な高さは倍率 M やフィルタの間隔 D などによって決まる。
例えば d=2.15μm（画素サイズの 1/2）, M=2, D=10mm, a=50mm とすれば、約 5μm の高さ測定が可能である。このように、式（3.8）によって最小測定高さや精度が決まる。
表 3.1 に対象物を自動ステージに載せて、高さを変化させた時のステージ高さと測定結果の差の分散を以下の表に示す。
表 3.1 視野の違いによる高さ測定値の標準偏差の例
視野

5mm

230μm

σ

5.5μm

6.1nm

・ 測定時間100ms 以下

・ 画素

1240×1024

・ メモリ

2GB

・ CPU

Pentium1.5GHz

3-14

補

足

（1）フィルター付マスクについて
フィルターを取り付ける窓の形状は円形として説明したが、図 3.A1 のように矩形でもよく、同じ形、同じ大きさであれば他の形状でも有効である。また、窓部の間隔は長い方が分離した像の間隔が大きくなる点では効果的であるが、光強度が弱くなってしまう。従って、間隔は対物レンズ径の半分程度が最適である。
なお、複数枚のレンズからなるレンズ系の場合、1 枚のフィルター付マスクは主面の間に
配置する。

Ｄ
マスク

赤 色 フ ィ ル タ ー
赤色フィルター

緑 色 フ ィ ル タ ー
緑色フィルター

図 3.A1 フィルター窓部の矩形の例（2）フィルターと照明光源
カラーCCD は、3 つの波長帯域に分離できる分光感度特性（RGB）を持っている。
フィルターはこの帯域に合致したものを用意する。さらに、照明用光源はこの帯域に十分な強度を有するものを使用する。また、色収差を考慮してフィルターの組み合わせは赤と緑または緑と青がよい。

3-15

関連用語の解説
バンプ
BGA 型パッケージなどにある白小球で IC チップとプリント基板とを接続するための微小なコネクタ。
ワイヤループIC チップの電極とリードフレームのインナーリードとを結んだ時の金線の形状。
ラインセンサ
主に CCD イメージセンサ素子、レンズ、ドライバーコントロール回路によって構成される。対象物の映像をレンズによって素子面に結像させ、光の量と位置から対象物の長さ等の諸元を測る。
レーザ変位計三角測量を応用した方式で、発光素子と光位置検出素子の組み合わせで構成されている。
測定対象物に照射された半導体レーザ光は対象物から拡散反射され、受光レンズを通して
光位置検出素子上にスポットを結ぶ。対象物が移動する毎に移動するスポット位置を検出することで対象物までの変位量を知ることができる。
白色干渉法
白色光源による干渉縞は、物体の参照面光路長差が０の場所のみ、白色の縞が観察される。参照鏡の位置を光軸に沿って移動させ、この白色縞ができる位置を求めることにより、物体の形状が測定できる。この手法を白色干渉法と呼ぶ。
ナイフエッジ法
ピント合わせの方法のひとつ。ごく薄いナイフの刃をレンズの焦点位置に置いて焦点像を遮断しその際の焦点像の変化によって焦点位置を判断する方法。
ピエゾ効果ある結晶に電圧を加えると電圧に比例してひずみが生じることを逆圧電効果と呼ばれる。
このような電圧効果・ピエゾ効果を生じる圧電性物質には、水晶、ロッシェル塩、チタン
酸バリウム、ジルコンチタン酸鉛などがある

参考文献〔1〕坂井一文：“分光位相差検出法による高さ測定”,産業技術総合研究所 特願 2008-203460

3-16

第4章

非合焦関数解析法による高さ測定

4.1

概

要

半導体や電子部品の外観検査では、部品の上側から観察される画像を用いてリード検査、
モールド検査、マーク検査などの 2 次元検査が行われている。他方、リードの浮きなどは高さ情報が必要になるため、側面観察やレーザ変位計などが用いられている。工程の効率化や品質保証の観点から、2 次元検査と 3 次元検査をインポケットの状態で検査されることが望ましいが、同時に検査できる装置はない。
検査に使われる撮像素子は、CCD や C-MOS が主流であるが、近年の撮像素子の発達によって画素サイズが小さくなり数百万画素の撮像素子が使われるようになって、これまで分離できなかった合焦点からのズレ（非合焦）の情報が得られるようになった。
本マニュアルは、従来の２次元検査の画像を解析し、焦点位置からのズレに対応した画像の広がり度合いを数値化し、微小な高さを測定する方法について解説する。
この装置は次のような特長を持つため、半導体プロセスの後工程への応用が期待できる。
①

被検査体の２次元検査用画像の強度分布からリードの高さを測定する。

②

一回の撮像により多数のリードの高さを測定することができる。

③

既存の撮影手段と画像解析ソフトウェアに高さを求める演算の解析ソフトを追加する
だけの簡単な形態である。

4-1

4.2

高さ測定の原理.

全ての結像光学系では、レンズの開口に伴う空間周波数の制限によって、1 点からの光は結像面では空間的な広がりをもっている。図 4.1 は 1 点(δ関数*)からの光がレンズを通り合焦点位置における光の強度分布を示すものである。この広がりは点応答関数(PSF)と呼ばれ、広がりの幅は光の波長λに比例し、レンズの開口数 NA*に反比例する。

1

強
度

-4

-2

0

2

4

μ m

図 4.1 合焦点位置における光の強度分布

また像の広がりは結像面(合焦点位置)からの距離に応じても変化する。図 4.2 はガウスビームがレンズにより収束する際の焦点位置近傍のガウスビームの広がりを示している。ここでｚは焦点からの距離、ω0 は焦点位置におけるビームの広がり半径を表わす。

z

ω0
z=0

図 4.2 焦点位置近傍のビームの広がり

4-2

図 4.3 は IC のリード先端部分を撮影した画像における強度分布を示す。焦点位置(ｚ＝0)
ではエッジからの光は実線のような強度分布を示すのに対し、焦点位置から前後にずれた
場合は破線のように強度が低下すると同時に強度分布の広がりが大きくなる。
強度
焦 点 位 置 （0）
ｚ＝

ピントずれ

図 4.3 リード先端の強度分布
レンズによるインコヒーレント*結像系では、像の強度分布 g は物体の形状関数 f と点応答関数（PSF 関数）h とのコンボルーション積分

g f

2

*h

2

（4.1）

  f ( x, y )h( x  x, y  y )dxdy (4.2)

によって与えられる。ｇ、ｆ、ｈのフーリエ変換をそれぞれＧ、Ｆ、Ｈとすると
(4.3)

Ｇ＝Ｆ・Ｈ
となる。

図 4.4 は IC のリード部分を拡大したものである。リードの高さをａ、リードの長さをｂとする。リードが本来の位置からｚだけ浮くことによりリード長さｂは⊿ｂだけ変化する。
ケ ー シ ン グ

リ ー ド

a
z
b
⊿b

図 4.4 IC のリード部分の拡大図
4-3

ここでリード先端部分が均一な輝度であると考えると図 4.5（ａ）のように b0＋⊿b の範
囲で矩形関数として表される。また点応答関数によりリード先端部分の強度分布は図4.5(b)のように広がりをもつようになる。

⊿b

⊿b

b0
(a)

b0
(b)

図 4.5 リード先端部分の矩形関数と像の強度分布

まず、式(4.3)からリード先端部分の強度分布のフーリエ変換Ｇ1 を求めると

G1  1  Az  B  22となる。但し、 A  2a 2
6b

 NA2
2

(4.4)

B

b02
3

 02

NA  0 / z 0

z 0   02 /  、  はフーリエ変換の座標である。ここでリード先端部分の画像における強度分布Ｑは

Q  1  G1 sin(2x0 ) max  ( Az  B) 2  02

(4.5)

で与えられる。但し、ξ0 はＱが最大値になる時の座標ξである。
これからＱとリード先端の高さｚとの関係をQ   Az  B  0  A1 z  B1
より、高さ z が求められる。

4-4

(4.6)

4.3

高さ測定の手順

非合焦関数解析法による高さ測定の手順を図 4.6 に示す。インポケットの IC はケース内
で様々な方向を向いているため、先ず、像の中心を画面の中心位置に移動後、回転操作（アフィン変換）を行う。基準ピン（図 4.6 では 1 と 12pin 及び 13 と 24pin）の先端位置で引いた直線を基準に各ピンの先端の強度分布を求め、原理で示した演算（フーリエ変換）によって各ピンの高さを計算する。
pin認識ﾏｰｸ
開

始

1

24

12

13

画像入力

画像1,12,13,24pinの先端中央位置を認識

像の中心を画面の
中心に移動

基準線

IC像を回転して平行にする

1と12pinおよび13と24pinの先端位置で直線を引く

直線を基準に各pinの先端の強度曲線を求める（各pinの幅で積算する）

フーリエ変換（虚部＊虚部）1/2
フーリエ変換（虚部）
0.7

波形の最大値 Imax を求める

0.6
0.5
0.4
0.3
0.2
0.1

高さ＝ k[1-Imax]1/2

0-0.1

Ｋは実測値より求める

終

-0.2
各pin

（注）虚部の1/2では負の値によってｴﾗｰになるため、虚部の二乗の1/4で
計算する。

了

図 4.6 高さ測定手順のフローチャート

4-5

4.4

実施例

図 4.7 に非合焦関数解析法によって IC リード（24pin）のコプラナリティ*を測定した例を示す。

図 4.7 上から撮影した IC 画像

図 4.8 は開発したソフトウェアの画面で、その結果を図 4.9 に示す。評価のため 6 ピンを故意に変形させている。

図 4.8 測定するソフトウェア

図 4.9

4-6

IC の各ピンの高さ

4.5

仕

様

図 4.10 は試料ステージを 0（基準点）から 100μm まで移動（5μm 間隔）しながら、非合焦関数解析法によってリード高さを測定する装置の概要を、図 4.11 にはその結果を示す。
図 4.11 より自動ステージの高さと測定値の標準偏差は 3.5μm であった。
（注）図 4.11 の直線はそれぞれのピンの高さを測定したものであるが、グラフ上で重なるのを防ぐため便宜上、切片の位置をずらしている。

C-MOS
ｶﾒﾗ

リング照明

Z軸移動量
Z軸移動量
制御装置
制御装置

自動Zｽﾃｰｼﾞ

図 4.10 ステージを使った測定装置

図 4.11 ステージの位置と測定の高さ【非合焦関数解析法による高さ測定仕様】
視野（mm)

10mm X 10mm

測定値の標準偏差（δ）

3.5 μm300ms 以下（Pin 位置認識、画像回転、平行移動の処理時間を含む）

・

測定時間

・

画素

1240×1024

・

メモリ

2GB

・

CPU

Pentium1.5GHz

4-7

補

足

ＭＴＦ（Modulation Transfer Function）
MTF は主にレンズの結像性能を評価するものである。 どのようにボケでいくか、ある
いは空間周波数によりどの程度ボケるのかを表す関数で横軸に細かさを表す空間周波数、縦軸に明るさの再現性を表すコントラストを定量化したものである。
MTF を求めるには PSF(点応答関数)をフーリエ変換する方法があり空間周波数とコント
ラストとの関係が連続で求められる。図 4.A1 は理想的な点光源(δ関数)をフーリエ変換したものである。この時、コントラスト応答は破線 A のように全空間周波数領域で一定の値を示しているのに対し、図 4.A2 のように PSF をフーリエ変換した MTF は実線Ｂのように
空間周波数が高くなるにつれコントラスト応答が弱くなっている。レンズを通し結像した像が焦点においてもボケるのは高い空間周波数領域の情報が失われていることを表している。さらにボケの強い PSF では破線Ｃのように低い空間周波数からすでにコントラスト応答が弱くなっている。

A
1

1

0.8
コ
ン
ト
ラ
ス
ト
応
答

0.8
コ
ン
ト
ラ
ス
ト
応
答

0.6
0.4
0.2

B

0.6
0.4

C

0.2

0
0

20

40

60

80

100

0

120

0

空間周波数 ＬＰ/mm

20

40

60

80

100

空間周波数 ＬＰ/mm

図 4.A1 点光源（δ関数）のフーリエ変換

図 4.A2 PSF のフーリエ変換

4-8

120

関連用語の解説
共焦点法
被写体を撮像するにあたり、レンズの位置を上下に動かし焦点が合った画像を複数枚撮影する。それらを画像処理することにより、物体表面すべてに焦点が合った画像を得る方法。
標準偏差
データの分布の広がりの度合いを表す。ｎ個のデータがあるとき、Ｘを相加平均とする
n

と標準偏差



 x  X 
k 1

2

k

n

但し、 X x1  x2 ・・・ x nnで表される。

δ(デルタ)関数ｘ≠0 の時、δ(x)=0 かつ  x dx  1 で定義される関数。


開口数（NA）
解像度を決める指数のこと。物体から対物レンズに入射する光線の光軸に対する最大角度をθ、物体とレンズの間の媒質の屈折率をｎとすると NA＝ｎsinθで表される。NA が大きくなると解像度の値が小さくなり細かなものが見えやすくなる。
空間周波数
空間的な周期を持つ構造の性質で、単位長に含まれる構造の繰り返しの多さを表す。
画像処理ではミリメートルあたりの線数をさす。
コントラスト応答
信号の最大値と最小値の差をコントラストといい、入力信号のコントラストに対する出力信号のコントラストの比をコントラスト応答という。
インコヒーレント波の持つ性質で、干渉のし易さを表し干渉縞が鮮明に出る状態をコヒーレントと言うのに対し、二つの波の振幅、位相が不規則に変化し干渉縞を作らない状態を言う。
コプラナリティ取付け面に対する部品の各端子や電極の最下面の均一性を表し、平坦度ともいう。

参考文献〔1〕坂井一文：“非合焦関数解析による高さ測定”,産業技術総合研究所 特願 2007-323120〔2〕MTF による光学機器の性能評価#3 MTF と PSF,
＜http://d.hatena.ne.jp/aquila2664/20081109#1226242651＞,(accessed 2009-11-09)

4-9

第5章

低コントラスト欠陥検出

5.1

概

要

外観検査において、異物、打痕及び変形等の正常部に対する特徴量が明瞭な欠陥に関しては、自動化が進んでいるが、しみ等に見られる正常部に対する特徴量が不明瞭な欠陥については、官能検査も含め様々な分野で、その自動検出方法が問題となっている〔1〕、〔2〕。
このことは、IC リードフレームにおいても同様であり、寸法検査や異物等の正常部に対する濃度差が明瞭な欠陥検査については、その自動化が行われているが、しみや光沢むら等の正常部に対する濃度差が不明瞭な欠陥(以下、低コントラスト欠陥)については、自動化が進んでいない。
本マニュアルでは、IC リードフレーム等で問題となっている低コントラスト欠陥に対して有効な検出システムの使い方について説明する。本マニュアルで示すシステムは、今まで検出が困難であった下記のような欠陥に対しても有効である。
(a)正常部との濃度差が 20 階調程度以下の欠陥。（ただし、フルスケール 256 階調の場合である。）(b)形状や大きさが一定しない。
(c)欠陥部の濃度に関する傾向が一定しない。(正常部と比べ、濃度階調が高い欠陥と低い欠陥の両方が存在する。)(d)正常品のばらつきが大きい。

5-1

5.2

コントラス欠陥検出手法の原理

5.2.1 現状の欠陥検出方法の問題点
通常、欠陥検出に用いられる手法として、検査画像と良品とされる参照画像間の濃度差による比較検査法〔3〕が用いられる。これは、良品を代表する標準サンプルと対応点毎に濃度に関する差の絶対値を計算し、あるしきい値以上の箇所を欠陥とする方法である(以下、
濃度差分法)。この手法は、処理がシンプルであり、その結果処理時間も短いため、一般に用いられているが、一つの良品サンプルとの比較では検出精度にばらつきが生じるため、検査対象毎に最適なしきい値を決定する手法〔4〕〔5〕や複数の良品サンプルから統計的にしきい値を決定する手法が提案されている〔6〕〔7〕〔8〕。しかし、IC リードフレームにおける低コントラスト欠陥では、製品自体の表面状態のばらつきや照明等の画像入力環境の変化により、統計的に求められた良品の濃度分布と欠陥部の濃度分布が重なってしまうた
め、これらの手法の適応は困難である。一方、低コントラスト欠陥検出方法として、欠陥モデルを用いた方法〔9〕が提案されているが、IC リードフレームにおける低コントラス欠陥は、良品部と欠陥部の濃度差が、良品の濃度にばらつきに対し、明瞭な差を期待できな
い場合、検出が困難となる。
図 5.1 は、欠陥部が異物などの高コントラス欠陥の場合の例を示しているが、良品の濃度のばらつきある場合、良品の誤検出を考慮したしきい値を設定することにより検出することが可能である。これに対して、低コントラスト欠陥の場合は、欠陥部と正常部との濃度差が小さいため、良品サンプルを元にした通常の濃度差分検出方法では、検出が困難と
なる。

5-2

5.2.2 本マニュアルで用いる検出手法の原理
本マニュアルにおける欠陥検出方法では、欠陥部の大局的な濃度変化に注目し、スプラ
イン関数＊〔10〕〔11〕の補間機能を用いて、検査対象ごとに擬似的な良品画像を生成し、その疑似良品画像と検査対象画像との濃度差分により欠陥検出を行う。また、疑似良品生成の際に、ラインプロファイル*上の濃度変化の大きさから判定した欠陥候補部の補間を行うが、本マニュアルで検出ターゲットとする低コントラスト欠陥は、異物等の欠陥と異なり、正常部と比較し明瞭な濃度差が無く、また、明度の高い欠陥と明度の低い欠陥が存在するため、正常部と比較し明るい箇所と暗い箇所のどちらを欠陥と見なして補間するか、
もしくは、その欠陥候補部を補間すべきかについては、局所的な情報だけでは判定できない。そのため、本マニュアルで示す手法では、遺伝的アルゴリズム(以下、GA)＊〔12〕を用いて、疑似良品構成面の全体的な滑らかさ等を考慮し、補間すべき箇所を判定している。
以上の処理により、図5.2において、検査画像から得られるラインプロファイルに、濃度変化の大きな箇所を正常部のデータを基に補間することにより、疑似良品画像を生成する(図5.2参照)。
このようにして構成した疑似良品画像は、尐なくとも良品エリアに関しては、検査対象
画像との濃度差を低く抑えることができるため、検出判定のためのしきい値を低く抑える
ことができる。これにより、正常部との濃度差が小さい低コントラスト欠陥検出が可能と
なる。

5-3

5.2.3 スプライン関数の使用方法
以下に、スプライン関数を用いた疑似良品画像作成方法について述べる。本マニュアルにおいては、スプライン関数としてリーゼンフェルトスプライン＊〔11〕を用いている。
スプライン関数の適応に際し、先ず、検査対象となるエリアに、図5.3に示すようにスプライン曲面を構成するためのサンプリング点(図5.3の黒丸。以下、データ点)を一定間隔に配置する。そして、y座標を固定したデータ点列から得られるスプライン曲線をt曲線とし、x座標を固定したデータ点列から得られたスプライン曲線をs曲線とする。ここで、今回採用したリーゼンフェルトスプラインでは、その端点部分は多重節点とし、固定点のように振る舞うため、検査領域の外周にダミーのデータ点(図5.3の白丸)を付加している。そして、ダミーのデータ点を合わせたデータ点の個数をx方向にI個、y方向にJ個とする。なお、付加したデータ点の濃度は、検査領域内の最近傍のデータ点と同じとし、データ点間の間隔も検査領域内のものと同じとする。
ここで、横方向i番目、縦方向j番目に位置するデータ点をサンプリングされた実数座標(xi,j,yi,j)及び濃度(zi,j)を要素とするベクトルPi,jとして、(5.1)式で表す。これにより、検査対象領域の濃度曲面は、(5.2)式に示すB-スプライン＊の積により求められる。なお、(5.2)
式において、K及びLは、各々t曲線及びs曲線の階数＊と呼ばれる値である。本システムでは、
K=L=4としている。

Pi , j  xi , j , yi , j , zi , j



(5.1)

(i  0, , I  1; j  0, ,J  1)
I 1 J 1

x (t ,s )   x i , j B i ,K (t )B j ,L (s )i 0 j 0

I 1 J 1y (t ,s )   y i , j B i ,K (t )B j ,L (s )(5.2)

i 0 j 0

I 1 J 1z (t ,s )   z i , j B i ,K (t )B j ,L (s )i 0 j 0

ただし、(0≦ｔ≦I-1)および(0≦s≦J-1)である。

5-4

x
t 曲線

y
0

1

i

I-2

I-1

1
データ点
pi,j

s 曲線
j
J-2
J-1

データ点間隔検査領域: データ点: 付加データ点

図5.3 ｓ曲線とｔ曲線

5-55.2.4 欠陥候補部の補間
ここで、5.2.3で示したs曲線及びt曲線における欠陥候補部の補間方法を示す。図5.4(a)は、s曲線もしくはt曲線上に観察される画像濃度ラインプロファイルの例を示している。
画像濃度ラインプロファイルは、通常、高周波成分のノイズを含んでいるため、スプライン関数のローパスフィルタ*としての機能を用い、低周波成分を抽出する(図5.4(b))。そして、このスプライン曲線上において、その振幅が後述する評価値(Qv)により、大きい箇所を欠陥候補として、正常部から推定した濃度で補間を行い、疑似良品面を生成する(図5.4(c))。
このようにして、作成された疑似良品面は、正常部では、正常面の濃度データの平均値となり、欠陥部のみ、画像濃度面と差異が生じるので、しきい値を小さくしても欠陥部のみ濃度抽出することができる。

座標(a) 画像濃度プロファイル
スプライン曲線

濃度

：データ点

座標
(b) 画像濃度プロファイルとスプライン関数

濃度補間で得られた疑似良品面

座標(C) 補間による疑似良品面の生成
図5.4 低コントラスト欠陥検出原理
5.2.5

モチーフの定義と合成処理

ここで、欠陥候補部の抽出方法の詳細を述べる。まず、照明等の撮影環境の影響を除去するため、各t曲線及びs曲線毎にシェーディング補正＊を行う。そして、シェーディング補正したスプライン曲線において、図5.5(a)に示すように、一対の極小点間に極大点が一つ存在する場合もしくは一対の極大点間に極小点が一つ存在する場合を調べ、これらをモチーフ(Motif)と呼ぶことにする(注：図5.5はt曲線を例に示している。)。そして、その両端部をモチーフの底点呼び、底点間に挟まれた極大点もしくは極小点をモチーフの頂点と呼5-6

ぶことにする。また、凸状のモチーフを山部、また、凹状のモチーフを谷部と呼ぶことにする。なお、t曲線に位置する検査領域の両端に位置するx1,jおよびxI-2,jについては、近傍の極点が極小の場合は極大点とし、また、極大の場合は極小点として取り扱うこととする。s曲線についても同様に処理する。そして、検出されたモチーフを検出された順にMν(ν=0,1,･･･)として、ラベルをつける。
次に、一つのスプライン曲線において、連続する谷部もしくは山部において、それらを含む大きな谷部もしくは山部が存在する場合は、小さなモチーフを合成し、新たな谷部もしくは山部として追加する。例えば、図5.5(a)の谷部Mν+3とMν+5の合成によりMν+8が谷部として追加される。
以上の処理を各スプライン曲線における全ての連続する谷部と山部について行い、初めに検出されたモチーフと合成により追加されたモチーフの合計をU個とする。

(a)欠陥候補モチーフの検出: 補間で得られたデータ点

濃度: 画像上のデータ点

補間候補エリア(C(r+1))ppε -1, j ε , jp(ε +d-1) , jp(ε +d), jp(I-1), j

p0 , jpε , j(Cr) p(ε +d-1), j(Cr)
補間候補エリア(C(r+2) )補間候補エリア(Cr )X軸(b)補間候補エリアの補間
図 5.5 モチーフの検出とその統合
5-7

5.2.6 補間候補の決定
5.2.5で検出されたモチーフについて、以下の判定基準により補間候補にすべきか否か決定する。先ず、5.2.5で検出されたモチーフMν(0 ≦ ν < U)について、底点間の距離(Lν)
をモチーフの幅とし、モチーフの頂点から底点間を結ぶ直線(Lineν)に降ろした垂線の長さ(Dν)をモチーフの深さとする(図5.5(a))。
次に、(5.3)式で示されるQνの値(これをモチーフ評価値と呼ぶこととする。)を調べ、あらかじめ設定したしきい値(Qth)以上のものを補間候補エリアCr(ただし、0 ≦ r < R-1; R≦ U)とする。Rは、Qν ≧ Qthのモチーフの個数である。%

Q ＝

Dν

(5.3)

Lν

(0  ＜U

ただし、U : 検出モチーフ数 )Lν：モチーフＭνの幅Dν：モチーフＭνの深さ
ここで、図5.5(b)に示すように補間候補エリアCrが得られた場合、その底点間に位置する
データ点(pε,j～p(ε+d-1),j)(1 ≦ ε ≦ I-2)について、それ以外のデータ点(p0,j～p(ε-1),jおよびp(ε+d),j～p(I-1),j)を元にCrエリアの補間を行い、新たなデータ点pε,j(Cr)～p(ε+d-1),j(Cr)を計算する。このようにして補間により求められたデータ点をpi,j(Cr)とし、その濃度をzi,j(Cr)とする。
なお、ここで求めた濃度zε,j(Cr)～z(ε+d-1),j(Cr)は、濃度面が滑らかとなる最適な組み合わせの選定時(5.2.8～5.2.13参照)に、補間候補エリアCrが選定された場合のデータ点pε,j～p(ε+d-1),jの濃度データとして用いる。

5.2.7 矛盾する補間候補対の決定
5.2.6で、得られた補間候補エリアにおいて、図5.5(b)におけるCrとC(r+1)あるいはC(r+1)とC(r+2)のように連続する山部と谷部あるいは谷部と山部が補間候補として選定された場合、これらの欠陥候補は、データ点の一部を共有しているため、同時に両者を補間部として選択することは矛盾する。そのため、このように他の欠陥候補とデータ点を共有する欠陥候補の組み合わせを「矛盾補間候補対」と呼ぶことにし、(r0(ω),r1(ω))(ただし、0 ≦ ω <Ω)として記録する。ここで、Ωは矛盾補間候補対の総数である。

5-8

5.2.8 遺伝的アルゴリズムの適応方法
5.2.6において選定された補間候補エリアは、tあるいはs曲線で判定されており、局所的な判定である。そのため、5.2.6で選定された補間候補エリアが疑似標準画像を構成する上で、補間すべきであるか否か、また、5.2.7で述べた矛盾補間候補対が存在する場合、どちらの補間候補を採用すべきかの判定は、大局的な判定を行う必要がある。そこで、本マニュアルで示す手法においては、GA (遺伝的アルゴリズム)を用いて、全体の滑らかさ等を考慮することにより、疑似標準画像を構成するために最適な補間エリアの組み合わせを選択する。なお、補間候補エリアが検出されなかった場合は、欠陥は存在しないものと判定する。
5.2.9 染色体の定義
本手法では、染色体として5.2.6で選定された補間候補エリアCrに対応するR個を設定する(図5.6)各個体の染色体は、対応する補間候補エリアに対応しており、1または0のいずれかの値をとる。そして、r番目の遺伝子の値をφ(r)とし、φ(r)=1のとき、補間候補エリアCrに含まれる補間されたデータ点の濃度値zi,j(Cr)(i=ε,･･･,ε+d-1)(図5.5(b)参照)を採用

0

補間エリア No.
r

1 0 0 1 0 0 1 0 1(R-1)

･･････ 0 1

図 5.6 染色体の定義
し、(5.2)式により疑似的な良品画像を計算する際のデータ点として用いることとする。
5.2.10

各データ点濃度の決定方法

データ点pi,jは、s曲線もしくはt曲線における補間候補により、複数の補間候補エリアに含まれる可能性がある。
ここで、データ点pi,jに対して、Hi,j個の補間候補エリアから濃度の推定があるとし、それらの補間候補エリアをCr(h)(0 ≦ h < Hi,j)で表すこととする。このときの、データ点pi,jの画像濃度z'i,jを以下のように決定する。先ず、Cr(h)の遺伝子状態が全て0のときは、サンプ
リングにより求められた画像濃度zi,jとし((5.4)式)、それ以外の場合は、遺伝子状態が1となっているCr(h)から推定される濃度の平均として表す((5.5)式)。

z i, j  z i , j(5.4)

その他の場合、5-9

H i , j 1

 z i j (C r h )(r (h ))z i, j 

h 0

,

( )

(5.5)

H i , j 1 (r (h ))h 05.2.11

適応度の定義GAを用いて、各個体の環境への適応度を以下の式により定義する。

F

1
1  G1 F1  G2 F2  G3 F3(5.6)

ここで、F1、F2およびF3は、次に示すペナルティであり、G1、G2およびG3は、各適応度の重み付けを決める係数である。
<適応度>
F1:補間候補エリアの中で、矛盾補間候補対が同時に補間対象となることに対するペナルテ
ィ。
F2:選定されたデータ点により構成されるスプライン面の滑らかさが正常部の条件から外れることに対するペナルティ。
F3:隣接データ点間の濃度差が正常部の範囲から外れることに対するペナルティ。
以下に、その詳細を述べる。
F1については、(5.7)式により、5.2.7で述べた矛盾補間候補対の内、どちらか片方の遺伝
子のみ1以外のときはペナルティを与えるようにする。

  1    r       r    

1

F1 

0

 0

1

(5.7)



ここで、r0(ω)とr1(ω)は、矛盾補間候補対であり、Ωは矛盾補間候補対の総数である。
また、記号  は排他的論理和*である。
F2については、先ず、GAで選択された補間候補エリアの組み合わせに基づき、5.2.10で示した方法により、全ての補間候補エリアに含まれるデータ点の濃度を計算する。次に、各欠陥候補エリア(Cr)(ただし、0 ≦ r < R)について、(5.3)式で示したモチーフ評価値Qr
を調べ、5.2.6で述べたしきい値Qth以上の場合、(5.8)式によりペナルティを与えるように
する。
R

(5.8)

VQ  r r

F2 ＝  0R

ただし、

1 Qr  Qthのとき 0 それ以外のとき

VQ (r ) ＝ 

5-10

F3については、検査領域内のデータ点pi,j(1 ≦ i ≦ I-2;1 ≦ j ≦ J-2)のうち、いずれかの補間候補エリアに含まれるデータ点の個数をNcとし、それらのデータ点をpi(c),j(c)(c=0,･･･,N_c)で表す。そして、pi(c),j(c)について、その4近傍のデータ点との濃度差を調べ、その値が5.3.1で述べるしきい値Zth以上であったとき、(5.9)式によりペナルティを与える
ようにする。
j (c )1
 i (c )1  i , j (c )   i (c ), j c 0  i i (c )1
j  j (c )1

F3 ＝
4N c
N c 1(5.9)

ただし、 i  i (c ) かつ j  j (c )1 z i(c ),j (c )  z i,j (c )  zth のとき i , j (c ) ＝ 
それ以外のとき01i (c ), j ＝ 0

5.2.12z i(c ), j (c )  z i(c ), j  zthのとき
それ以外のとき

GAの適応度の定義

GAにおける次世代の個体の生成方法として、各個体の適合度に比例して確率的に個体を選ぶルーレット戦略＊〔12〕を用いた。また、優秀な個体を次世代に存続させるエリート保存戦略＊〔12〕を併用した。
先ず、世代毎の個体の中で適合度Fの高い順からS%の個体を優秀個体として、次世代にそのまま生存させる。そして、(100-S/100)%の個体については、ルーレット戦略で選択した個体により交叉と突然変異＊を行い、次世代の個体を決定するものとする。
5.2.13GAの進化方法と終了判定

交叉は、二つの個体間で染色体を組み替えることによって新しい個体を生成するものであるが、本論文においては2点交叉*を用いた。即ち、ルーレット戦略で選択された親個体Ａ、Ｂについて、ランダムに選択した染色体上の2点間の遺伝子を親個体Ａ、Ｂ間で入れ替えて、新たな２つの子供個体を生成する。そして、これらの子供個体を次世代の個体とする。また、交叉によりできた子供個体について、  %の確率で突然変異を発生させている。突然変異を行う個体については、ランダムに遺伝子を1個選択し、そのビットを反転させている。
以上の処理を所定の個体数が揃うまで繰り返し、次世代の個体を決定する。
上記の方法によりGA処理を行い、各世代において最大適応度をもつ個体の適合度を記録する。そして、その値が指定の世代数の間変化しない場合(デフォルト値: 15世代)、もし
くは最大適合度が設定したしきい値(デフォルト値:0.9)以上となったとき、GAの処理を終了し、最大適合度を有する個体の組み合わせを、疑似良品面を構成するための補間を実行
する補間エリアの組み合わせとする。
5-11

5.2.14判

定5.2.13で得られた補間エリアの組み合わせに基づき、(5.2)式により疑似良品面を構成し検査対象画像と濃度差分を行う。そして、設定したしきい値(Ith)以上の箇所を欠陥部とし
ている。

5.3

低コントラスト欠陥検出性能評価実験

本検出手法の性能評価を行うため、以下の実験を行った。
5.3.1 パラメータの説明
本検出システムを利用する場合は、以下のパラメータを設定する必要がある。
＜検査対象に関するパラメータ＞(a)Qth：補間候補決定のためのしきい値。5.2.6で述べたモチーフの評価値Qvがこの値以上の時、補間候補対象となる。(デフォルト値：0.1)＜スプライン関数に関するパラメータ＞(a)データ点間隔：5.2.3で述べたスプライン関数を構成するためのデータ点間の間隔を設定する。データ点間隔を小さくすると、微細な濃度変化の検出が可能となるが、処理時間がかかる。また、データ点間隔を広くすると、画像濃度が全体的に平均化されてしまい、
欠陥部の濃度変化をうまく捉えることができない。本システムでは、デフォルトで10画素
間隔としている。
＜遺伝的アルゴリズムに関するパラメータ＞(a)G1： F1に対する重み付け。(デフォルト値：0.5)
注）F1:補間候補エリアの中で、矛盾補間候補対が同時に補間対象となることに対す
るペナルティ。
(b)G2：F2に対する重み付け。(デフォルト値：0.3)
注）F2：選定されたデータ点により構成されるスプライン面の滑らかさが正常部の条件から外れることに対するペナルティ。
(c)G3：F3に対する重み付け。(デフォルト値：0.2)
注）F3：隣接データ点間の濃度差が正常部の範囲から外れることに対するペナルティ。
(d)Zth：F3に関するしきい値。近傍との濃度差がこれ以上の場合、ペナルティが与えられる。
(デフォルト値：10階調)(e) GA終了しきい値：(5.6)式の評価値（F）がこの値以上となった場合、GA処理を中止し、そのとき最大適合度を有する個体のパターンを、疑似良品面構成のための補間の組み合わせとする。(デフォルト値：0.9)
5-12

(f) GA終了世代数：ここで、指定する世代数において、最大適合度の値が変化しない場合、GA処理を終了する。(デフォルト値：15世代)＜判定に関するパラメータ＞
Ith：補間候補決定のためのしきい値。5.2.6で述べたモチーフの評価値Qvがこの値以上の時、補間候補対象となる。(デフォルト値：6階調)
5.3.2 性能評価実験条件
画像入力部はカメラ、照明装置からなり、検査画像の入力を行う。照明は、ドーム照明、
リング照明があり(図5.7)、それ以外にも疑似同軸落射照明等の照明も可能であり、今回の実験では、疑似同軸落射照明を用いた。図5.7で使用している機器は、以下の通りである。
(a)カメラ：東芝テリー製

CS3960DCL

撮像素子：2/3型インターライン方式CCD*
有効画素数：1392(H) x 1040(V) 正方格子配列
走査方式：プログレッシブスキャン＊
映像出力：カメラリンク＊出力準拠

データ 12/10/8bit切換

レンズマウント：Cマウント(b)ドーム照明：CCS製

HPD-150-SW(白色光)

電源：24V/20W(c)リング照明：CCS製

LDR2-90BL(青色)

電源：24V/15W
検査画像は、解像度640×480画像のモノクロ画像を用いている。前処理として、3×3の平滑化フィルタ処理〔15〕によるノイズ除去を行い、正規化相関法〔3〕によるパターンマ
ッチングにより、位置合わせを行った。また、各手法とも設定したしきい値以上の箇所を欠陥として検出するが、検出誤差等を考慮し半径直径2画素(約100μm)以下のものは削除することとした。なお、使用したコンピュータのCPUクロック数は、2.16GHzである。なお、
5.3.1に示したパラメータについては、デフォルト値を用いた。
ＩＣリードフレーム部のパッド部分を検査対象として、検査員により、良品と判定された
サンプル72個及び欠陥と判定されたサンプル20個について、提案手法と矢口らによる平滑スプライン関数を用いた欠陥検出方法〔13〕〔14〕(以下、平滑スプライン法)および背景
別統計的しきい値法〔6〕による手法を比較した。検査範囲(図5.3参照)は、幅130画素及び縦150画素(実エリア：8mm×9.2mm)である。また、欠陥については、正常部との最大濃度差に応じ、(a)20階調未満の場合、(b)20階調以上40階調未満、及び(c)40階調以上の3種類に分けて性能評価を行った。

5-13

5.3.3 性能評価実験結果
図5.8に本マニュアルに示す検出方法で検出実験を行った結果を示す。図5.8(a)、(b)共に、
低コントラスト欠陥部を検出している。また、比較のために、参考文献〔13〕〔14〕の平滑化スプラインを用いた方法と文献〔6〕の背景別統計的しきい値を用いた欠陥検出例を表5.1に示す。本マニュアルで示す検出手法は、良品の誤検出はあるものの、全体的な検出精
度は最も良い結果が得られている。本手法の平均処理時間に251msec.であった。

カメラ

ドーム照明

リング照明

図 5.7 画像入力部

図 5.8 しみ欠陥検出例
5-14

表 5.1 欠陥検出比較実験

検査
サンプル

正解判定率
正常部との濃度差:ID
サンプル数 本マニュア 平滑化スプ 背景別統
計的しきい
(階調)
ルの手法
ライン法
値法
ID<20

8個

75%

37.50%

0%

20≦ID<40

10個

100%

100%

50%

40≦ID

2個

100%

100%

100%

良品サンプル

72個

93.10%

66.70%

100%

良品サンプル

92個

92.40%

68.50%

85.90%

251msec.

240msec.

31msec.

欠陥サンプル

平均処理時間5-15

5.4

ICリードフレーム検査システム

5.4.1

ICリードフレーム検査システムの概要

本欠陥検出システムの外観を図5.9に示す。Xステージ部にICリードフレームを載せ、パソコンから操作することにより、ステージが移動し、Xステージ部に載せたICリードフレー
ムがラインセンサカメラの下を通過することにより、画像が１ラインずつ順次取り込まれ、
制御パソコンでそれを合成し、ICリードフレーム1枚の画像を構成する。照明は疑似同軸落
射照明を用いている。取り込まれた画像は、制御パソコンでパッド毎に欠陥検査が行われる。図5.10-1及び図5.8-2に制御パソコンに制御画面を示す。図5.10-1(a)が初期画面であり、
ラインセンサ調整、ステージコントロール、画像入力及び検査実行の項目からなる。各ボ
タンスイッチの機能は、以下のとおりである。
＜画像入力＞
・図5.10-1(b)の画面により画像入力方法をラインセンサカメラもしくは既に登録した画像
ファイルからの入力を選択し、試験的な画像入力を行う。取り込んだ画像は保存可能である。位置決めパターンの設定等に用いる。
＜ステージコントロール＞
・ステージに関する前進、後退及び停止のマニュアル操作を行う。また、原点検出機能があり、検査実行を行う前に、必ず原点検出を行う必要がある。

5-16

（ａ） 検査装置全体写真（ｂ）画像取り込み部
図 5.9 検査システム外観
5-17

（b）画像データ入力画面（ａ）メイン画面（c）ステージ操作画面（d）ラインセンサ調整画面
図 5.10-1
(1)

5-18

操作画面

図 5.10-2 操作画面(2)＜ラインセンサ調整＞

・ラインセンサカメラの調整を行う。具体的には、「フォーカス調整」でピント、「ゲイン調整」で感度及び「タイミング調整」で照明及びラインセンサカメラへのパルス出力タ
イミングの調整を行う。
＜検査実行＞
・ICリードフレームの検査を行う。図5.10-2に示す画面が表示されるので、「パラメータ調整」により、パッド位置検出パターンの設定及び下記(5.3.1)に示すパラメータの設定を行う。
そして、「検査実行」をクリックすることにより、ステージが移動し、画像入力及び検査が行われる。入力された画像は、「全体画像」部に表示され、欠陥が見つかった箇所については、赤枠で囲まれ番号が割り当てられる。そして、「欠陥表示部」に欠陥部の拡大写真が表示され、欠陥部が赤丸で示される。

5-19

5.4.2 欠陥検出実験例
本検査システムを用いた欠陥検出例として、図 5.11 にしみ等の低コントラスト欠陥検出例及び

図 5.12 に異物等の高コントラスト欠陥検出例を示す。

図 5.11

低コントラスト欠陥検出例

5-20

図 5.12 高コントラスト欠陥検出例

5-21

5.5

仕

様

(a)ラインセンサカメラ：日本エレクトロセンサリデバイス(株)製 NUCL4KDmk2
画素数：4096 画素
画素サイズ：9.325μm(H)×9.325μm(V)
素子長：38.1925mm
データレート：40MHz(固定)レンズマウント：ニコンマウント
エンコーダ：光洋電子製 TRD-SH2000B(2000 パルス/回転)(b)X ステージ：シグマ光機(株)製 SGSP46-500(X)
移動量：500mm
最大移動量：120mm/sec
位置決め精度：0.025mm
繰り返し位置決め精度：0.006mm
2 軸ステージコントローラ：SHOT-202(ｼｸﾞﾏ光機製)(c)照明：シーシーエス(株)製

疑似同軸落射照明

LFV2-130RD

発光色：赤色
照射エリアサイズ：100mm×100mm
消費電力：12V/28W(d)ステージ制御パソコン：DELL 製 Precision T3400
CPU:375W

インテル(R)Core(TM)2 Quad プロセッサ

Q9559(12MB L2 キャッシュ、2.83GHz、1333MHz FSB)(e)画像処理ライブラリ：Matrox 製

MIL9.05-22補足（欠陥解析）各欠陥の特徴に関する解析結果を述べる。下記の写真は，マイクロスコープで各欠陥を観察したものである。図 5.13～5.15 に欠陥の例を示す。マイクロスコープは、キーエンス製
VHX-1000 を用いた。なお、照明は、同軸落射照明を用いている。
図 5.13 は、しみ欠陥であり、大きさや正常部との濃度差も様々である。(a),(b)の例では、比較的はっきりとした小さなしみであるが、(c),(d)の例では、広範囲の変色が見られる。
図 5.14 の光沢ムラでは、欠陥部が不明瞭であり、大きさも広範囲に広がっている。また、
図 5.15 に打痕欠陥の例を示す。図 5.15(a)は上方から見た写真であり、図 5.15(b)は、それを 3 次元表示したものである。図 5.16 に異物欠陥の例を示す。図 5.15 と同様に、(a)に上方から見た写真、(b)にその 3 次元表示画面を示す。

図 5.13 しみ欠陥例5-23図 5.14 光沢ムラ欠陥例

図 5.15 打痕欠陥例
5-24

図 5.16 異物欠陥例

5-25

関連用語の解説
スプライン関数「スプライン」という言葉は、製図のとき滑らかな曲線を描く道具(自在定規)を意味している。スプライン関数で曲線を当てはめることは雲形定規や自在定規によって曲線を描く
かわりに数式的に行っていることに当たる［10］。スプライン関数は、サンプリングされた
データ点から、計算により曲線や曲面を当てはめるものである。
ラインプロファイル
画像上に設定された直線上における濃度変化を示したもの。
遺伝的アルゴリズム(GA)
生物進化（選択淘汰・突然変異）の原理に着想を得たアルゴリズムであり、確率的探索・
学習・最適化の一手法と考えることができる〔12〕。GA を用いることにより、高速に最適化を行うことができる。
リーゼンフェルトスプライン
通常のスプライン曲線は、与えられたデータ点を補間するものであるが、リーゼンフェ
ルトスプラインでは、データ点を制御点として働くスプラインであり、他のスプラインと異なり、連立方程式を解く必要がないという特徴がある。
B-スプライン種々のスプライン関数を構成するうえで、基礎(basis)となるスプラインであり、スプラ
イン関数は一般的に B-スプライン関数の重ね合わせとして表すことができる。
階数
スプライン関数の滑らかさの指標となるものであり、この値が大きくなるとスプライン

曲線及び曲面は滑らかさを増す。
ローパスフィルタ
信号に含まれる各周波成分のうち、低周波成分のみを通過させるフィルタのことである。
シェーディング補正
画像入力を行う際に、照明状態、撮像素子の特性等の影響などによって入力画像に生じた濃度レベルのむら（シェーディング）を補正すること。
排他的論理和論理演算の一つであり、共に真、共に偽のときに、偽を返す。その真と偽が異なるときに、真を返す。
ルーレット戦略
GA において、次世代の個体を決定するための元となる親を選ぶ方式であり、各個体の適応度に応じた確率により、次世代の個体を作るための親(2 個体)を選択する。適応度に応じた個体が、選択されやすくなる。

5-26

エリート保存戦略
GA において、次世代の個体を決定する方法のひとつであり、適応度の高い個体をそのま
ま次世代に残す方法である。
突然変異
GA において、染色体上のある遺伝子座の値を他の対立状態の値に置き換えることにより、新しい個体を生成するものである。
2 点交叉
GA において、染色体上の 2 点で挟まれた範囲を個体間で交換することにより、新しい個体を生成するものである。
インターライン方式 CCD
受光部で光電変換した信号電荷を一斉に垂直レジスタに移したあと、垂直転送 CCD に移し、その後、水平転送 CCD から１ラインずつ読み出す CCD です。
CCD
CCD（Charge Coupled Device）は、光として入射された画像信号を電気信号に変換する機能を有した素子のことで、入射された光を光電変換により電荷に変換し、その電荷を外部に転送します。
プログレッシブスキャン
ノンインターレーススキャンとも呼ばれ、インターレーススキャンが一本置きに画像走査を行うのに対して、この方式は上から順次走査を行います。
カメラリンク
カメラと画像信号取り込み部とのインターフェース規格の一つです。
ラインセンサカメラ
通常のエリア状に CCD が配置された通常のカメラと異なり、CCD が 1 列に並んだカメラです。通常のカメラより、高解像度の画像を得ることができますが、画像入力に際しては、ステージと組み合わせて、画像を１ラインずつ入力する必要があります。

5-27

参考文献
〔1〕秦清治:”見えにくい欠陥・評価しにくい欠陥・人の感覚に依存する欠陥の外観検査”，
電気学会産業システム情報化研究会資料,(2006), pp.23-26.
〔2〕中野宏毅,森由美:”マルチウェーブレットを用いた低コントラスト欠陥の定量評価手法 液晶ディスプレイのムラの抽出・定量化手法”,View2004 ビジョン技術の実利用ワーク
ショップ，横浜(2004), pp.44-47.
〔3〕 (社)精密工学会画像応用技術専門委員会編:”画像処理応用システム”,東京電機大学出版局(2000),p.98,pp.208-209.
〔4〕前田俊二,窪田仁志,牧平坦:”LSIウェーハパターン自動外観検査における最適しきい値の自動生成”,信学論誌D-ⅡJ79 No.2 (1996-2), pp.191-201.
〔5〕前田俊二,遠藤文昭,牧平坦,窪田仁志:”比較検査における数値解析手法を用いたしきい値設定支援”,信学論誌D-ⅡJ81 No.1 (1996-2), pp.27-36.
〔6〕浜松玲,渋谷久恵,西山英利,大島良正,前田俊二,野口稔:”背景別統計的しきい値法を用いた半導体ウェハ検査技術”,View2004 ビジョン技術の実利用ワークショップ，横浜(2004), pp.1-5.
〔7〕沼田宗敏,輿水大和:”階層化パラメータを用いた学習型大局的欠陥検査”，動画像処
理実利用化ワークショップ2008，(2008), pp.222-227.
〔8〕岩田健司,佐藤雄隆,尾崎竜史,坂上勝彦:”逐次更新型統計的リーチ特徴によるロバス
ト背景差分”,View2008ビジョン技術の実利用ワークショップ,横浜(2008), pp.294-299.
〔9〕石井明,越智保雄,阿久津正幸:”欠陥モデルによる内部溶接欠陥の自動認識”，日本機械学会論文集（Ａ編），60巻578号（194-10)，No.94-0335，(1994),pp.272-277.
〔10〕桜井明：”スプライン関数入門”，東京電機大学出版局(1981).
〔11〕桜井明，菅野敬祐，吉村和美，高山文雄：”Cによるスプライン関数”，東京電機大学出版局(1993).
〔12〕北野宏明："遺伝的アルゴリズム"，産業図書(1993),p.138,p.13,pp.136-137,〔13〕小野豊土，矢口博之，市野学:”液晶パネルの自動検査システムの基礎的研究"，電子情報通信学会情報・システムソサイエティ大会，(1997),ｐ171.
〔14〕渡部典生，市野学，矢口博之：”むら検査方法および装置”，公開番号：特開平11-66311（1999).
〔15〕高木幹雄，下田陽久：”新編 画像解析ハンドブック”,東京大学出版会(2004),pp.1899-1900.

5-28

第６章

スペックル干渉法を利用した
３Ｄ変形分布計測

6.1

概

要

半導体パッケージは複数の熱膨張の異なる部材で構成された複合材料である。そのため、実装時や通電時（作動時）発熱による温度変化により、部材の熱膨張の違いに由来する動作不良や破損などの不具合が生じる場合がある。そこで電子部品の信頼性向上のために半導体製品の実装時や通電時（作動時）の高精度な２Ｄ－３Ｄ変形計測が求められている。
実装時や通電時（作動時）の変形を把握することにより、動作不良や破損などの不具合が生じないパッケージの構造設計、材料選定が可能となる。この章では３次元の変位分布計
測が可能なスペックル干渉法について紹介し、駆動させた時の IC パッケージ変形挙動計測
例を紹介する。
レーザ干渉を利用した計測法はレーザの波長程度の計測分解能を持つ高精度な計測手法である。よく知られているマイケルソン型＊やフィゾー型＊の干渉計では平坦な面を持つ参照面との比較により形状分布（高さ）を計測する面外方向の計測法である。また、モアレ
干渉法は、測定物表面に回折格子を貼り付け、表面が変形することで回折格子が歪み、歪んだ様子が干渉縞として得られる手法で、面内方向の変形分布を計測する手法である。一方、今回用いるスペックル干渉法は、面外と面内の両方の変形分布計測が可能な手法である。スペックル干渉法はレーザ光が粗面などによって散乱・反射することで生じるスペックルパターンを利用した計測手法で，実構造物のような粗面を持つ測定物の計測法として非常に適している。測定領域は理論的には顕微鏡領域から数 m2 程度の広範囲までの変位分布計測が可能である。計測分解能はレーザ干渉の位相解析を行うことで数 nm～数 10nm となり、非常に高精度な計測法である。

6-1

6.2

スペックル干渉法

6.2.1 スペックル干渉法の原理[1,2]
本研究で利用したスペックル干渉法の原理について説明する。レーザ光を粗面に照射した場合、レーザのコヒーレンシ（可干渉性）によって ランダムな位相の光が重ね合い、図6.1 に示すように干渉し、ぎらぎらと輝く斑点模様が観察される。この干渉パターンはスペックルパターンと呼ばれる。

図 6.1 スペックルパターン
スペックル干渉法は、２つ以上の光路を通ったレーザ光を重ね合わせた時に生じるスペッ
クパターンを利用して変位を求める方法で、図 6.2 に面外変位分布を計測する光学系の一例を示す。この光学系は１つのレーザ光をハーフミラーにより２方向に分け、測定物と参照物の表面で散乱・反射させる。両方の表面で散乱・反射したレーザ光を測定面（CCD カメラ）上で結像させてスペックルパターンを得る。ここで測定面が面外方向に ｗだけ変形したとすると、変形前後で二つの光路を通る光の位相差Δθは次式で表される。

 

4w

(6.1)



ここで、λはレーザの波長である。また、測定物と参照物からの反射光が CCD カメラセン
サの１画素上に到達する光の振幅をそれぞれ a1，a2 とし、両者の位相、並びに位相差をそれぞれ1,2, =1-2 とするとこの点の変形前の強度は次式で表される。

6-2

参照面
測定物

レーザ光

変位 w
レンズ
ハーフミラー
絞り

CCD

図 6.2 スペックル干渉法による面外変位計測用の光学系の模式図

I before  a1 exp(i1 )  a 2 exp (i 2 )

2

(6.2)

 a12  a 22  2a1 a 2 cos
変形後の強度 Iafter は変形による位相差Δθを用いると
I after  a12  a 22  2a1 a 2 cos(   )

(6.3)

となる。
次に変形前 Ibefore と後 Iafter のスペックルパターンの差の絶対値を取るとI before  I after  2a1 a 2 [cos( )  cos(   )] 4 a1 a 2 sin( ) sin( )
2
2

(6.4)

となり、この式より|sin(+Δθ/2)|は 0 から 1 までランダムな値をとり、|sin(Δθ/2)|
は 0 から 1 まで規則的に変化することが分かる。変形による位相差Δθがπの奇数倍(2n+1)πの場合は、|sin(Δθ/2)|=1 となり強度が最も大きくなり、偶数倍 2πn の場合は、|sin(Δθ/2)|=0 となり強度がゼロとなる。そのため表面に沿った変形分布が連続的に変化している場合、明暗の干渉縞として変形分布が形成される。この干渉縞１本当たりの変位量は Δ

θ＝２πとし、式(6.1)よりレーザの波長の半分となる。この原理を利用し、縞分布よりレーザ照射方向の面外変位分布を求めることができる。
一方、面内変位に関しては図 6.3 に示す光学系のように測定面の法線に対して対称な２方向からレーザ光を照射し、法線方向から観察するとレーザの照射方向、即ちｘ軸方向の面内変位分布を求めることができる。測定面が面内方向に u だけ変形したとすると変形前
6-3

レーザ光
測定物
CCD

変位 u

δ

レンズ

δ
X

絞り
レーザ 光

Z

図 6.3 スペックル干渉法による面内変位計測用の光学系の模式図

後で二つの光路を通る光の位相差Δθは次式で表される。
 

4u



sin 

(6.5)

ここで、λはレーザの波長で、δ は照射光が測定面の法線となす角である。面内変位により生じた干渉縞は縞１本当たりの変位量はΔθ＝２πとし、式(6.5)より求められ、縞分布よりレーザ照射方向の面内変位分布が得られる。
6.2.2 スペック干渉法の数値化
スペックル干渉法で得られる干渉縞一本当たりの変形量は数 100nm 程度であるが、干渉縞を数値化することで更に高精度なナノレベルでの計測が可能となる。数値化の方法として広く使われている方法に位相シフト法がある[1]。この方法は面外変位の場合は参照用のレーザ光、面内変位（２光束法）の場合は片方のレーザ光の位相を変化させたものを３枚以上取り込み、演算し、数学的に位相を求める方法である。この方法では３枚以上の画像を取り込むため基本的に静的（もしくは準静的）な条件での計測に限られているが、高精度な位相解析が可能である。しかしながら実現象（熱膨張やクリープ現象＊、応力緩和＊など）は時々刻々と変化しているため動的な変形挙動計測が重要である。次に、動的変形挙動計測法を紹介する。
動的な計測法はスペックルパターン画像中の１画素の強度変化に着目し、強度変化を時系列に評価し変形を求める。本計測法では時系列評価法で変形方向を同定するために一方向変位キャリアを利用した位相解析法を用いる[2,3]。面外変位計測で参照面を一方向に、一定速度 u で移動させながらスペックルパターンを計測した場合、１画素の時系列強度変6-4intensity (a.u.)

250
200
150
100
50
0
0

1

2

3

4

5

time (s)図 6.4

参照面が速度(u=1000nm/s)で移動している時の１画素の時系列周期変化

intensity (a.u.)

250
200
150
100
50
0
0

1

2

3

4

5

time (s)図 6.5 参照面が速度(u=1000nm/s)で移動し、測定物が速度(v=400nm/s)で移動している時の１画
素の時系列周期変化

16basemeasurement

phase ()

12
8
4
0
0

1

2

3

4

5

time (s)図 6.6

図 6.4（base）と 6.5（measurement）の位相接続結果

6-5

化は図 6.4 のような一定周期の強度変化（基準変位）となる。一方、測定物の変形（一定速度で変形が生じていると仮定）が付加されると変形方向に応じて図 6.5 に示すように図6.4 の周期が変化する。図 6.4,5 の周期変化を位相に変換し、位相接続した結果を図 6.6 に示す。破線は基準変位で、実線は変形が付加されたものである。両者とも変形条件が一定のため一定の傾きで位相が時間とともに増加している。ここで両者の差の絶対値（単位：π）に定数（この場合はλ／４）を掛けたものが変形量となり、測定物、参照面ともにカメラに近づく向きを正とすると、基準変位キャリアのグラフより下側にある場合は、参照
面の変形方向と測定物の変形方向は同じで、基準変位キャリアのグラフより上側にある場
合は逆方向の変形となる。例えば、基準変位キャリアがカメラに近づいている場合、図 6.6の場合では測定物はカメラの方向に向かって変形していることになる。

6-6

6.3

スペックル干渉法を利用した変形分布計測装置の使用方法

計測準備①装置を起動させて、画像をモニターに表示させる。
②モニターを見ながら測定対象物と装置の位置やレンズを調整し、最適な大きさに測定物を映し出す。
③レーザを起動させ、レーザの出力とレンズの絞りを調整しながら最適なスペックルパタ
ーンを発生させる。この際、スペックルパターンの輝度値がカメラの階調値の中に入るようにする。
（白すぎたり、黒すぎたりしない）④微動ステージを動作させ、モニター上のスペックルパターンが明暗を繰り返す様子を観察し、再度明るさの調整を行う。
⑤スペックル干渉縞が得られていることを確認するために、微動ステージを動作させながら５秒ほど画像を取り込む。解析ソフトを立ち上げ、干渉縞作成ボタンで画像を読み込
み、スペックル干渉縞を作成し、コントラストの良い干渉縞が得られているか確認する。
もし、干渉縞のコントラストが不十分な場合は、再度③～⑤を繰り返す。
本計測⑥IC の駆動時の変形挙動を計測する場合は、最初に微動ステージを動作させた後、画像の取り込みを行い、続いて IC を通電させる。
⑦取り込みを終了し、IC の通電を止め、微動ステージを止める。
⑧解析ソフトで画像を取り込み。この際、２の N 乗の数だけ画像を取り込む。（例えば256,512,1024 枚）⑨画像中で解析を行う位置を決めて、解析を行う。
⑩解析結果は BMP と CSV 形式のファイルとし出力される。

6-7

6.4

実施例

6.4.1 実施例 1
IC チップを駆動させて５秒後の面外変位分布計測結果（１）スペックル干渉縞

白枠部の解析結果（解析点：61x91）

解析結果（3x3 平均化処理）

解析結果（3x3 平均化処理、３次曲面式近似）
6-8

IC チップを駆動させて５秒後の面外変位分布計測結果（２）スペックル干渉縞

白枠部の解析結果（解析点：61x91）

解析結果（3x3 平均化処理）

解析結果（3x3 平均化処理、３次曲面式近似）
6-9

IC チップを駆動させて５秒後の面外変位分布計測結果（３）スペックル干渉縞

白枠部の解析結果（解析点：61x91）

解析結果（3x3 平均化処理）

解析結果（3x3 平均化処理、３次曲面式近似）6-10

6.4.2 実施例 2

IC(SN74LS04N)の写真

6-11

IC チップ（SN74LS04N）を駆動させて５秒後の面外変位分布計測結果（4）スペックル干渉縞

単位：nm
800

600-800600

400-600400

200-400
200

0-200

0
-200

（3x3 メディア処理と平均化処理）
6-12

系列1

191

181

151

171

白枠部の解析結果（解析点：191x61）

系列13
161

141

131

111

系列25
121

101

81

系列37
91

71

51

61

31

系列49

41

11

21

1

系列61

-200-0IC チップ（SN74LS04N）を駆動させて 10 秒後の面外変位分布計測結果（5）スペックル干渉縞

単位：nm
1,200-1,500
1,500

900-1,200

1,200

600-900

900
600

300-600

300

0-300

（3x3 メディア処理と平均化処理）
6-13

系列1
190

172

163

145

181

白枠部の解析結果（解析点：191x61）

系列11
154

127

136

109

系列21
118

91

100

73

82

55

系列31
64

46

28

系列41

37

19

-300
1

系列61
系列51
10

0

-300-0IC チップ（SN74LS04N）を駆動させて 5 後の面内変位（水平方向成分）分布計測結果（6）スペックル干渉縞

単位：nm
1,200

800-1,200

800

400-800

400

0-400

0

-400-0-400-800--400

-800
系列66
191

181

171

161

151

141

131

121

111

91

81

71

51

41

31

21

1

系列1

11

系列14

61

系列27

101

系列40

-1,200--800201-1,200

系列53

白枠部の解析結果（解析点：201x66）（3x3 メディア処理と平均化処理）
6-14

IC チップ（SN74LS04N）を駆動させて 10 後の面内変位（水平方向成分）分布計測結果（7）スペックル干渉縞

単位：nm
1,200-1,800
1,800

600-1,200
1,200

0-600
600
0

-600-0-600-1,200--600-1,200-1,800--1,200-1,800-2,400--1,800

191

181

171

161

141

131

121

91

101

81

71

51

41

31

21

1

系列1

11

系列15

61

系列29

111

系列43

151

系列57

201

-2,400

白枠部の解析結果（解析点：201x66）（3x3 メディア処理と平均化処理）
6-15

IC チップ（SN74LS04N）を駆動させて 5 後の面内変位（垂直方向成分）分布計測結果（8）スペックル干渉縞

単位：nm
1,200

800-1,200
800

400-800
400

0-400
0

-400-0-400-800--400-800
系列1-1,200--800

系列19
1

10

19

28

46

55

64

73

82

91

109

118

127

136

145

154

163

181

190

199

系列73

172

系列55

100

系列37

37

-1,200

白枠部の解析結果（解析点：199x69）（3x3 メディア処理と平均化処理）
6-16

IC チップ（SN74LS04N）を駆動させて 10 後の面内変位（垂直方向成分）分布計測結果（9）スペックル干渉縞

単位：nm
1,800

1,200-1,800
1,200

600-1,200
600

0-600
0

-600

-600-0-1,200-1,200--600-1,800-1,800--1,200

系列1
1

10

19

28

46

55

64

73

82

91

109

118

127

136

145

154

163

181

190

199

系列73

172

系列55

100

系列37

37

系列19

白枠部の解析結果（解析点：199x69）（3x3 メディア処理と平均化処理）
6-17

6.4

仕

様

図 6.9 面外計測時の写真

図 6.7 面内計測用ヘッドの写真

図 6.8

装置の主な仕様
・使用するレーザの波長：633nm，635nm, 690nm
・空間分解能：640x480pixels
・画像取り込み速度：60, 30, 15, 7.5 fps
・測定範囲：最大 20x20mm
・連続計測時間：10 分以上（HDD の容量に依存）

6-18

コントローラの写真

関連用語の解説
マイケルソン型
アメリカの物理学者マイケルソンによって考案された二光線干渉計で、光速度の測定に用いられたことで有名である。
http://100.yahoo.co.jp/detail/%E3%83%9E%E3%82%A4%E3%82%B1%E3%83%AB%E3%82%BD%E3%8
3%B3%E5%B9%B2%E6%B8%89%E8%A8%88/
フィゾー型
フィゾー型干渉計は、測定物表面の極めて近い位置に基準平面（極めて平行度の高い平面板）を配置し、レーザ光を測定物に照射し、基準平面と測定物表面からの反射光を干渉させ、測定物表面の面精度を非接触で測定する方法のこと。
クリープ現象クリープとは、一定の温度、一定の応力を受ける材料が、ある時間を過ぎた後に生じる変形のこと。
http://www.instron.jp/wa/resourcecenter/glossaryterm.aspx?ID=28
応力緩和
一定温度で長時間にわたり一定のひずみを呈している材料において応力が減尐すること。
http://www.instron.jp/wa/resourcecenter/glossaryterm.aspx?ID=150
３Ｘ３平均化
画像の縦横３列３行の輝度値を使用して平均値を求める処理のこと。
３次曲面式近似f(x,y)=Ax3+By3+Cx2y+Dxy2+Ex2+Fy2+Gxy+Hx+Iy+J の３次曲面式を利用した近似処理
のこと。
参考文献
[1] Pramod, K. R., Digital Speckle Pattern Interferometry and Related Techniques,John Wiley & Sons, Ltd Press, Chichester (2001).
[2] 井口学他、実験力学ハンドブック、朝倉書店、p.185-188（2008）.
[3] Uchino, M., Development of Dynamic Deformation Measurement Method Using Electronic Speckle Pattern
Interferometry,

International Conference on Advanced Technology in Experimental Mechanics 2007,

OS12-1-3(CD), (2007).
6-19

第7章

ま と

め

九州地域の牽引産業と位置付けられている半導体産業の内、多くの中堅･中小企業の展開する半導体後工程の検査技術を対象とし、九州産業技術イノベーション創出共同体形成事
業・研究開発環境支援事業「高精度・高スループット 2D-3D インスベクション技術の開発」（平成 20 年度～21 年度、経済産業省）に取り組んだ。本事業は、現場からのニーズが特に
強 い、IC リードフレーム検査、ワイアボンディング検査、半導体パッケージ検査、およびプリント基板(実装品)検査を中心に、企業の技術課題解決に資する試験・評価・分析方法を確立、それをマニュアル化し、企業への技術移転の促進および企業技術者等の技術力向上に資することを目的としている。本事業にはこれらの技術課題に関連した各地域企業への対応実績および技術ポテンシヤルを有する、産総研九州センタ－、福岡県工業技術センタ－、熊本県産業技術センタ－および鹿児島県工業技術センタ－の 4 研究機関が参画した。
平成 20 年度には画像処理のみにより精度の高い位置、濃淡、高さ情報を取得するための技術である「焦点深度の深い鮮明な撮像法」、「低コントラスト欠陥検出法」、「２Ｄ画像に基づく３Ｄ情報取得法」、および「高精度２Ｄ‐３Ｄ変形測定法」について、原理実証装置の試作を中心にマニュアル化に取り組んだ。上記４研究機関の共同体研究開発事業として各課題を分担し、５回にわたる検討会を開催し、各研究機関相互の連携･調整を図りつつ、本事業を推進し、次の成果を得た。
「焦点深度の深い鮮明な撮像法」では、半導体パッケージ及びプリント基板を対象として、多焦点撮像光学系の構築、照明系の構築、画像合成ソフトの開発を行い２焦点及び３焦点撮像法、照明系、画像合成ソフトに関するマニュアルを作成した。
「低コントラスト欠陥検出法」では、IC リードフレームにおけるしみ等の欠陥特徴の解析と欠陥検出ソフトの開発を行い、低コントラスト欠陥検出ソフトに関するマニュアルを作成した。
「２Ｄ画像に基づく３Ｄ情報の取得法」では、非合焦関数解析法について、画像解析ソフトの開発とこれを用いた QFP 型パッケージのリード浮き、パッケージ下面・上面の接地
面からの高さ、傾き等の検出性能評価、およびプリント基板を対象としたリード浮きの検査性能評価を実施し、これらの諸元を検出する技術マニュアルを作成した。さらに、分光
位相差検出法については、画像解析ソフトの開発を行い、高さ測定の画像解析ソフトについてマニュアルを同様に作成した。
「２Ｄ‐３Ｄ変形測定法」では、スペックル干渉法を利用した動的 3 次元変形計測法による半導体パッケージの面内および面外変形計測の実証試験を行い、マニュアルを作成した。
平成 21 年度にはそれぞれの課題について、生産現場に対応した応用装置の試作、実証試
験に取り組み、各検査技術の確立を図ると伴に、マニュアルを統合･完成させた。半導体パ
ッケージ、プリント基板およびワイヤ・ボンディングについて、画像処理のみによる様々な 2D-3D 検査項目を高速で精度良く測定し、かつ IC リードフレームのしみ、傷、付着する異物などの認識精度が高く、不良品の発生を大きく低減させることが可能な「高精度・7-1高スループッド 2D-3D インスペクション技術」を確立している。
なお、平成 21 年度後半からは、本マニュアルの成果普及を促進するために、九州地域内の半導体製造関連企業を主な対象として、各地で合計 4 回の成果普及発表会（画像処理技術研究会講演会）を開催した。各課題についての成果報告をはじめ、大学等から招へいした専門家による特別講演を含み、試作装置の展示･実演を行った。本発表会の参加者は、以下の通りである。
第1回

平成 21 年 11 月 16 日

産総研九州センター

32 名（7 社、16 名）

第2回

平成 21 年 12 月 8 日

鹿児島県工業技術センター

第3回

平成 22 年 1 月 21 日

KKR 熊本

第4回

平成 22 年 2 月 18 日

福岡県工業技術センター機械電子研究所

37 名（10 社、22 名）

37 名（10 社、20 名）
33 名（12 社、

18 名）
参加者合計

139 名（39 社、76 名）

ここで、カッコ内の数値は、参加者の内訳として、民間企業からの参加者を示している。
本事業は、平成 21 年度末で終了であるが、今回形成された産総研および公設研との公的研究機関のネットワークを中核として、今後も半導体産業を中心とした九州地域の民間企業群との連携を深めていく計画である。さらに、本技術は、半導体組み立て工程における
２Ｄ－３Ｄインスペクションのみならず、電子材料や各種精密部品など、他の生産現場における様々な計測･検査技術等に広く波及させることができる。画像処理応用分野においては、これまで有機的な繋がりに乏しかった産学官の研究者・技術者ネットワークを新たに形成し、将来の新技術開発につながる活動の場が構築できる。今後も企業等に対する本マ
ニュアルの普及を進め、技術移転･人材育成に寄与すると共に、この分野の新たな技術ニーズの収集に努める。ネットワークとしてこれに対するソリューションの提供に取り組むことで地域からのイノベーション創出につなげていく所存である。
最後に、本事業を推進するに当たり、ご支援･ご協力をいただいた関係各位に心からお礼を申し上げる。
平成 22 年 2 月末日

7-2

【問い合わせ先一覧】【第 2 章、第 3 章、第 4 章、その他全体的内容】
独立行政法人

産業技術総合研究所 九州センター〒841-0052 佐賀県鳥栖市宿町 807-1
TEL. 0942-81-3592

FAX. 0942-81-4089

http://unit.aist.go.jp/collab-pro/ci/wholesgk/index.html【第 2 章、第 3 章、第 4 章各章の対象物がプリント基板】
鹿児島県工業技術センター〒899-5105 鹿児島県霧島市隼人町小田 1445-1
TEL. 0995-43-5111.

FAX. 0995-64-2111

http://www.kagoshima-it.go.jp/【第 5 章低コントラスト欠陥検出】

熊本県産業技術センター

ものづくり室重森〒862-0901 熊本県熊本市東町 3-11-38
TEL. 096-368-2101

FAX.096-365-5704

http://www.kmt-iri.go.jp/【第 6 章スッペックル干渉法を利用した 3Ｄ変形分計測】
福岡県工業技術センター

機械電子研究所〒807-0831 福岡県北九州市八幡西区則松 3-6-1
TEL. 093-691-0260

FAX. 093-691-0252

http://www.fitc.pref.fukuoka.jp/center/meri/meri.htm

高精度・高スループット
２Ｄ‐３Ｄインスペクション技術マニュアル（第 8 章機器操作マニュアル）

平成２１年度九州産業技術イノベーション創出共同体形成事業
研究開発環境支援事業「高精度・高スループット２Ｄ－３Ｄインスペクション技術の開発」

目

次

第8章

機器操作マニュアル··············································································· 頁

8.1 産総研保有機器·························································································8-1
8.2 レーザー変位計·························································································8-3
8.2.1 準備 ··································································································8-3
8.2.2 測定 ··································································································8-3
8.2.3 データの保存 ······················································································8-3
8.2.4 終了 ··································································································8-4
8.3 分光光度計·······························································································8-5
8.3.1 準備 ··································································································8-5
8.3.2 測定 ··································································································8-5
8.3.3 終了 ··································································································8-6
8.4 ビジョンプロセッサ···················································································8-7
8.4.1 準備 ··································································································8-7
8.4.2 検査 ··································································································8-7
8.4.3 終了 ··································································································8-7
8.5 高速マイクロスコープ················································································8-8
8.4.1 準備 ··································································································8-8
8.4.2 測定 ··································································································8-8
8.4.2 動画の保存 ·························································································8-8
8.4.3 終了 ··································································································8-9
8.6 微小高さ制御装置···················································································· 8-10
8.4.1 準備 ································································································ 8-10
8.4.2 高さ制御 ·························································································· 8-10
8.4.2 終了 ································································································ 8-10
参考文献 ······································································································· 8-11

第8章

機器操作マニュアル

8.1

産総研保有機器

レーザー変位計

分光光度計

ビジョンプロセッサ

高速マイクロスコープ
8-1

微小高さ制御装置

8-2

8.2

レーザー変位計

8.2.1 準備①

電源確認を行う。（OA タップの電源スイッチ）②

測定ヘッド（LT-9010M）、ステージ（KS-1100）およびパソコン（PC）の電源ス
イッチを入れる。

③

PC デスクトップ上の”KS-Measure”を起動させる。

④”KS-Measure”操作ウインドウの上部にある「接続確認」アイコンをクリックする。

⑤

原点サーチ実行を促すダイアログが出てくるので「はい」を選択する。→ステー
ジが初期化される。

⑥”KS-Measure”操作ウインドウ右下部のステージの操作領域にある操作切り替え
で「PC」アイコンをクリックする。

8.2.2 測定①

対象物をステージ上に設置する。

②

測定ヘッドの高さを調整：LT-9010M の測定レンジ（±300μm）を考慮してモニタ画面の高さ表示バーで確認しながらヘッドを最適な高さに調整する。

③

測定ヘッドを設定する。
（ヘッドのプログラム 0 に基本的な条件を設定している。
その他の詳細設定は LT-9000 シリーズのユーザーズマニュアル参照）④ ”KS-Measure”操作ウインドウ右下部の「ステージの操作」領域にある移動アイコンで対象物をステージの適切な位置へ移動する。（通常は測定範囲の左上）⑤

プログラムの設定を行う。（測定範囲、測定ピッチ、移動速度等の設定）⑥

測定を実行する。（「プログラム測定開始」アイコンをクリック）

8.2.3 データの保存①

測定終了後「保存」アイコンをクリックすると、「名前を付けて保存」ダイアログが表示されるので、ファイル名を入力して「保存」をクリックする。
（”KS-Analyzer”
で読み出すことができるファイル形式（拡張子：md1）で保存させる）②

CSV 形式での保存は「ファイル」メニューの「CSV 保存」をクリックすると、「名前を付けて保存」ダイアログが表示されるので、ファイル名を入力して「保存」
をクリックする。

8-3

”KS-Measure”のメイン画面

8.2.4 終了①

メイン画面の右上にある「」アイコンをクリックして”KS-Measure”を終了させ、Windows も終了させる。
②

測定ヘッド（LT-9010M）およびステージ（KS-1100）の電源スイッチを切る。

③

OA タップの電源スイッチを切る。
以上

8-4

8.3

分光光度計

8.3.1 準備①

電源確認後（OA タップ電源スイッチ）、PC の電源スイッチを入れる。

②

PC デスクトップ上の”BWSpec”を起動させる。

8.3.2 測定①

分光光度計本体の受光部を対象光源の方向に向け、「」アイコンをクリックしてスペクトルデータを取得する（受光部表面を傷つけないように注意のこと）。
“BWSpec”のメイン画面※データがオーバーフローまたは微弱な場合、積算時間を調整すること。

8-5

②

測定終了後「」アイコンをクリックすると、「名前を付けて保存」ダイアロ

グが表示されるので、ファイル名を入力して「保存」をクリックする。（”Pixel
Format TXT File (*.txt)”または”Wavelength Format TXT File (*.txt)”で保存させる）③ Excel ファイルにエクスポートする場合には「」アイコンをクリックすると、

Excel データが自動的に起動する。
8.3.3 終了④

メイン画面の右上にある「」アイコンをクリックして” BWSpec”を終了させ、Windows も終了させる。
⑤

OA タップの電源スイッチを切る。

8-6

8.4

ビジョンプロセッサ

8.4.1 準備①

電源確認（OA タップ電源スイッチ）②各装置の電源を入れる。
（ビジョンプロセッサー、ステージコントローラー、照明）③

ステージのインジケーターで「RDY」を確認④

ステージの FUNCTION ボタンを押す。（「PO２」表示）⑤

モニタの「ツール」メニューから「カメラ表示」を選択するとキャプチャー画面が表示される。

⑥「クロス表示」ボタンをクリックするとクロスラインが表示される。

⑦

ステージ操作ボックスの「原点」ボタンを押し、原点復帰させる。

⑧

ステージ操作ボックスの「X」,「Y」ボタンにより、クロスラインの中心に IC の中心を移動させて「位置セット」ボタンを押す。

⑨

キャプチャー画面の「終了」ボタンをクリックする。

⑩「検査設定」メニューの「検査モードへ」を選択する。

8.4.2 検査①

ステージ操作ボックスの「START」ボタンを押し検査を開始する。

②

NG のときは「STOP」ボタンを押して「START」ボタンの点滅を解除させてから、再度「START」ボタンを押して検査を継続させる。

♦

照明のスペクトルを分光光度計によりモニタリングしながら検査することにより、照明条件とエラーモードとの相関を確認することができる。

♦

目的により分光位相差検出用のカメラや高速度カメラを設置して高さ情報や振動情報を計測することが可能である。

8.4.3 終了①

検査モード画面の左下にある「ティーチングモードへ」ボタンをクリックする。

②「ビジョン」メニューの「シャットダウン」を選択する。

③「VISION の終了」ダイアログの「電源を切れる状態にする」にチェックして「OK」
をクリックする。

④

OA タップの電源スイッチを切る。

8-7

8.5

高速マイクロスコープ

8.5.1 準備①

電源確認（OA タップ電源スイッチ）②

前面パネルの POWER スイッチ「③メイン画面が表示されたら「簡単録画」を選択する。

④

レンズの選択をする。（その他のレンズユニット）⑤

多焦点撮像装置の設定を行う。（ピント、あおり等）⑥

画面に表示されるガイダンスに従い、撮影条件の設定を行う。（明るさ、フレーム

」を押す。

レート等）
8.5.2 測定①

録画枚数を設定する。

②

録画方法を設定する。
「停止ボタンを押すまで録画し続ける」チェックボックス；
オフ（スタートトリガ：「停止」ボタンをクリックする前に録画枚数に達したとき、その時点で停止）オン（エンドトリガ：「停止」ボタンをクリックする前に録画枚数に達しても停止せず、「停止」ボタンをクリックすると、その時点までの動画が録画される。（古いデータは順次消去される））③「録画開始」ボタンをクリックして、録画を開始する。（ボタンの表示は「停止」
に替わる）

④

「停止」ボタンをクリックして、録画を終了する。（プレビュー保存画面が表示される）

8.5.3 動画の保存①

プレビュー保存画面で動画を再生しながら、保存を開始するフレームで一時停止
し、「始点セット」ボタンをクリックする。

②

保存を終了するフレームで一時停止し、「終点セット」ボタンをクリックする。

③「データ保存」ボタンをクリックすると「録画データ保存」ダイアログが表示されるので保存条件（ファイル形式、フォルダ、ファイル名）を設定する。

④「保存」ボタンをクリックし、データを保存する。

8-8

8.5.4 終了①

プレビュー保存画面で「終了」ボタンをクリックすると「簡単録画の終了」ダイ
アログが表示されるので「電源を OFF する」をクリックする。

②

OA タップの電源スイッチを切る。

補足
動画エディター（Movie Editor v1.2 KEYENCE）は、備え付け PC にインストールされている。必要に応じて解析、編集を行うことができる。
高速カメラで取得した画像を USB メモリーなどで備え付け PC にコピーする。PC 起動後、上記 Editor アイコンをクリックし、編集する動画を指定する。
① コマ画像の取得② 画質の修正（コントラスト、輝度）③ 動画の切り取り④ fps の変更
⑤ 画像保存形式の変更が可能である。編集した動画は、各自必要に応じて各自のメディアに save すること（産総研では、動画の保管は行っていない）。また、編集用 PC は他動画（外観検査関連画像に限定する）の編集用としても使用可能である。

8-9

8.6

微小高さ制御装置

8.6.1 準備①

電源確認（OA タップ電源スイッチ）② 各 機 器 の 電 源 ス イ ッ チ を 入 れ る 。（ PC 、 パ ル ス ス テ ー ジ コ ン ト ロ ー ラー：”SHOT-602”、ピエゾステージコントローラー：”FINE-503”）③ PC デスクトップ上の”SG Commander”を起動させる。
8.6.2 高さ制御①

表を参照し目的に応じて機種・接続先・軸番号を設定する。

② 「機種の選択」画面で機種をチェックし「次へ>>」ボタンをクリックする。
③ 「通信設定」画面の接続先を選択し「次へ>>」ボタンをクリックする。
④ 「接続テスト」画面の「テスト実施」ボタンをクリックして結果”OK”を確認した
後、「完了」ボタンをクリックする。
⑤ ”SG Commander”のメイン画面にある「ステージ操作」ボタンをクリックする。
⑥ ”Axis Control” 画 面 の 軸 番 号 を 選 択 し ”Move” あ る い は ”Move While

MouseDown”ボタンでステージを駆動させる。
動作内容と機種・接続ポート・軸番号の関係
動作

X 軸(粗動)Y 軸(粗動)Z 軸(粗動)X 軸(微動)Y 軸(微動)Z 軸(微動)機種

SHOT-602

SHOT-602

SHOT-602

FINE-503

FINE-503

FINE-503

接続先

COM4

COM4

COM3

COM1

COM1

COM1

軸番号

1

2

1

1

2

3

最小移動量
X,Y 粗動軸・・・

2µm

Z 粗動軸

1µm

・・・

X,Y,Z 微動軸・・・10nm（クローズド制御時）
8.6.3 終了①

”Axis Control”画面の「閉じる」ボタンをクリックする。

② ”SG Commander”のメイン画面にある「終了」ボタンをクリックする。
③ Windows も終了させる。
④

OA タップの電源スイッチを切る。

補足
8-10

本ステージは高精度（nm オーダー）の制御が可能であるため、試料（IC など）の固定は重要である。試料固定として① スペーサー② L アングル③ フレキシブルアングル④ ポール、ポールホルダー⑤ クロスアングルが用意されている。必要に応じて使用可能である。使用後は、元の位置（箱）に戻すこと。
尚、産総研では試料の保管業務は行っていないので、使用後は各自持ち帰ること。

高精度・高スループット２Ｄ‐３Ｄインスペクション技術マニュアル改訂表

8-11