Product Datasheet

Safetyspace™
Filter Tips
Filter Tip to Protectfrom Contamination

Product Information

Description

Filter tips are an efficient way to avoid cross-contaminationof your sample and/or pipette and reduce the need forpipette maintenance. Sartorius Safetyspace™ Filter Tips havemore space between the sample and the filter than conventional filter tips, which prevents the risk of the sample being
absorbed into the filter. Safetyspace™ Filter Tips are ideal
for use with Sartorius pipettes, enabling maximum accuracy
and precision.

Safetyspace™ Filter Tips are made of virgin polypropyleneand the filter of polyethylene without any self-sealing additives to avoid contamination of the sample, which may affectthe results. All the steps in the manufacturing process strictly
follow ISO 9001 and ISO 13485 quality standards, and thetips are manufactured in ISO 8-classified cleanroom conditions to avoid any contamination. Every production batch ofSafetyspace™ Filter Tips is tested for DNase, RNase, human
DNA and endotoxins. The tips are released for sale once they
have fulfilled the test requirements. Certificates are availableat www.sartorius.com.

The additional space between the sample and the filter
differentiates Safetyspace™ Filter Tips from other filter tips.
This feature is particularly useful in reverse pipetting,pipetting foaming liquids (such as buffers and proteins), or
when using Multi-Dispensing mode in electronic pipette.

-----Features
Available in volumes from 10 to 5,000 μL
All production batches certified DNase, RNase, human
DNA and endotoxin free
Informative rack labels indicating tip volume, productnumber, lot number, date, and purity level for improved tip
identification and traceability
All Safetyspace™ Filter Tips are E-beam pre-sterilized
Perfect fitting and sealing with all Sartorius pipettes
Smooth and ergonomic tip attachment and ejection
ensured by compatibility with the spring-loaded Optiload
mechanism of Sartorius pipettes
Color-coded tip trays make it easy to match tip withappropriate Sartorius pipette
Available in Single Tray flip-top racks that are easy to open
and close with one handMaterials used in tips, trays, and racks are 100 % recyclable
or can be incinerated as energy waste

Safetyspace™ Filter Tips are also available as Low Retention
tips. The Sartorius tip offering also includes Optifit nonfiltered tips and Optifit Low Retention non-filtered tips (seecatalog for details).

2----Applications
Molecular biology
Microbiology
Cell culture applications
Radioactive work
Any application where preventing cross-contamination
is vital

Intended Use
The Sartorius Safetyspace™ Filter Tips are intended, designed
and manufactured to be used in liquid dispensing in avariety of applications and to be used in combination withSartorius pipettes.
It is recommended that Safetyspace™ Filter Tips are usedwith Sartorius pipettes to ensure optimum compatibility
and performance.Technical Data
Technical Specifications*
Sterilization

Purity certificate

Endotoxin purity testing

DNase purity testing

Materials and Recylability
Electron beam (beta irradiation) in accordance with EN 552 and ISO 11137 andvalidated by using 25 kGy as the minimum
dose with SAL (sterility assurance level)of 10-6.

Tip

Polypropylene (PP)
Recyclable, can be incinerated asenergy waste

Polyethylene (PE)Polyethylene (PE)Recyclable, can be incinerated as energy
waste

Tray and Rack

Validated test result with Chromogenic
kinetic method according to Ph.Eur. 2.6.14,
Method D for endotoxin free pipette tip is< 0.005 IU/mL (EU/mL).

Polypropylene (PP)
Recyclable, can be incinerated asenergy waste

Cardboard package

Cardboard
Recyclable, renewable waste

DNase-free pipette tips show no evidenceof DNase activity in the fluorometric assay
with the detection level of < 6.25*10-5 U/μL
when DNase I is used as a standard.

Package Options

Every production batch (lot number) oftips is certified free of DNase, RNase,human DNA and endotoxins.

RNase purity testing

RNase-free pipette tips show no evidenceof RNase activity in the fluorometric assay
with the detection level of < 3.125*10-9 U/μL
when RNase A is used as a standard.

Human DNA purity testing

Human DNA free pipette tips show no evidence of human DNA in quantitative PCR(qPCR) using human DNA specific primers
and with a detection level of < 1 pg/μl.

Standards

ISO 9001 Quality management standard
ISO 14001 Environmental standard
ISO 13485 Medical device quality manage­
ment standard
ISO 14644-1 Class 8 cleanroom manu–facturing standard

Autoclavability

NoTransportation
conditions

Temperature: –50 °C to +50 °C
Humidity: 10 % – 90 %Shocks: < 10 G

Storage conditions

Temperature: –20 °C to +40 °C
Humidity: 10 % – 60 %Filter pore size

20 – 40 µm

Expiry date

5 years from manufacturing date*

Single Tray racks
purity-certified tip racks* Extended filter tip sterility assurance level, DNase, RNase, human DNA and endotoxin testing
detection limits may differ from values reported here. Please request a batch specific certificate
from lhinfo.finland@sartorius.com

3Ordering Information
Safetyspace™ Filter Tips
Order CodeVolume Range [µL]

Length [mm]Package

Purity Level
Certified free of DNase,
RNase, human DNA,
endotoxins

790011F

0.1 – 1031.5

Single Tray

783201*

0.1 – 10 Extended

46

Single Tray

790021F

0.5 – 2051

Single Tray

790101F

2 – 12051

Single Tray

790201F

5 – 200

52.5

Single TrayLH-XF780201

5 – 200 Extended

77.5

Single Tray

790301F

5 – 300

52.5

Single Tray

791001F

50 – 1,00078Single TrayLH-XF781001

10 – 1,000 Extended

102

Single Tray

791211F

50 – 1,20090

Single Tray

LH-795001F

100 – 5,000150

Single Tray

* Filter tip with regular air gap

Germany
Sartorius Lab Instruments GmbH & Co. KG
Otto-Brenner-Strasse 2037079 Goettingen
Phone +49 551 308 0F or further information, visit
www.sartorius.com

Specifications subject to change without notice.
Copyright Sartorius Lab Instruments GmbH & Co. KG.
Status: 10 | 2020

USASartorius Corporation
565 Johnson Avenue
Bohemia, NY 11716
Phone +1 631 254 4249
Toll-free +1 800 635 2906

-Tips/Unit
Pre-sterilized-Finland & Baltics
Sartorius
Biohit Liquid Handling Oy
Laippatie 100880 Helsinki
Phone +358.9.755.951
Fax +358.9.755.95.292
lhinfo.finland@sartorius.com

10 x 9610 x 9610 x 9610 x 9610 x 9610 x 9610 x 9610 x 968 x 9610 x 961 x 50