State-of-the-art model with body renewed sophisticatedly
Pursuing easy-operation, attractive new functions
have been implemented!

Rotary Evaporator

N-1300E·V·S Series

EYELA as a reliable brand name
Rotary Evaporator N-1300E·V·S Series

N-1300E-W

N-1300V-W

N-1300S-W■ Composition & Specifications

Product nameComposition

Rotary Evaporator

N-1300 main unit
driving part

E·V·S
glass set

N-1300 main unit
driving part

E·V·S
glass set

Free selection of N-1300
main unit driving part,
E·V·S condenser, wateror oil bath.

Water bathw/o bath
−Bath type
Bath temp. control range & accuracy

Water bath
RT + 5 ～ 90˚C ± 1˚C

E·V·S
glass set

N-1300 main unit
driving part
Water/oil bath

Water/oil bath
RT + 5 ～ 180˚C ± 1.5˚C ( oil ±3˚C)

E type glass set suitable for installation in fume hood.

E type condenser: Vertical double helix condenser with built-in adapter (cooling area 0.117m2)
Rotary joint: 29/38, ID 18xL 178mm
Sample flask (Eggplant shape): 1L 29/38
Receiving flask (Round bottom shape): 1L Ball joint S35/20

V type glass set applicable to both small or large volume flask.

E·V·S glass set

E type condenser

V type condenser

S type condenser

V type condenser: Vertical double helix condenser with built-in adapter (cooling area 0.146m2)
Rotary joint: 29/38, ID 18x L 178mm
Sample flask (Eggplant shape): 1L 29/38
Receiving flask (Round bottom shape): 1L Ball joint S35/20

S type glass set applicable to both low or high boiling point liquid.

Rotary joint

ModelCat. No. for 230V, 50/60Hz
Cat. No. for 115V, 60Hz
Rotation speed
Evaporation capacityRotation speed setting & display
Jack function
MotorHeaterVacuum seal
Bath inner dimensions (mm)Bath material & capacity
Bath inlet terminal
Bath connection nozzle
Ambient temperature
Dimensions(Max. height) (mm)Power source

N-1300E
266492266499Sample flask(Eggplant shape)

N-1300V
266432266439Receiving flask(Round bottom shape)

N-1300S
266372266379S type condenser: Diagonal double helix condenser (cooling area 0.146m2)
Rotary joint: 29/38, ID 18x L 272mm
Sample flask (Eggplant shape): 1L 29/38
Receiving flask (Round bottom shape): 1L Ball joint S35/20

N-1300E-W
266512266519N-1300V-W
N-1300S-W N-1300E-WB N-1300V-WB N-1300S-WB
26645226639226653226647226641226645926639926653926647926641910～310rpm
Max. 23mL/min (Water evaporation)
Setting by dial Digital display
Manual balancing system (Jack stroke 180mm, stepless)DC blushless motor−
1.05kW
1kW
Vacuum seal (Teflon® +Teflon® · Viton double seal) 1 set
Genuin parts: vacuum seal 2 sets Cat. No. 142610−ID 220 x 120H
ID 240 x 120H−SUS 304 4.3L
Aluminum (Teflon coating) 5L
For connection to evaporator main unit driving part Max. 2A
Cooling hose nozzle · Suction nozzle OD 10mm
5～35˚C
E: 514W x 342D x 645(825)H 8.8kg
E: 578W x 352D x 645(825)H 12.7kg
E: 565W x 352D x 645(825)H 13.3kg
V: 497W x 342D x 823(1003)H 8.9kg
V: 543W x 352D x 823(1003)H 12.8kg
V: 531W x 352D x 823(1003)H 13.4kg
S: 672W x 342D x 504(684)H 8.2kg
S: 736W x 352D x 504(684)H 12.1kg
S: 724W x 352D x 504(684)H 12.7kg
126VA · AC115V/253VA · AC230V, 50/60Hz
1.1kVA · AC115V/2.6kVA · AC230V, 50/60Hz
1.1kVA · AC115V/2.5kVA · AC230V, 50/60Hz* Performance data was taken under20°C of ambient condition with rated power and voltage.
* Adjustable accuracy of bath temperature was taken value when a sample flask was turning.
* Performance of evaporation differs depending on revolution speed, vacuum condition, bath temperature, condensing temperature, sample flask.
* There is F series (EYELA COAT®) which is applied with coating on glassware and superior with chemical resistance, transparency, cold resistance (-80°C) and heat resistance (120°C).

“Renewal design from the conventional evaporator”

Differences of New type evaporator, N-1300
It is renewal as N-1300 after 50 years since we had launched our firstevaporator N-1 and our past successive evaporators have continued to be well
accepted and highly appraised at laboratories over half century. Design isrenewed from the conventional model and it has finished up as a product which
can constitute better laboratory environment.

High flexibility in installation and Capability in space efficiency
Possible to install glassware set
from either of Right or Left handside to fit in installation spot.

A vertical E-type condenser suitedfor fume hood
A vertical condenser with a built-in adapter(E type condenser) has compact design inconsideration with using in fume hood.
Moreover, the condenser has been designedwith no-reverse flow from capillary without
having any concern, and with efficient vapor
collection in spite of the compact size.

Glassware can be set at either of right orleft hand side of machine body. A machine
can be set up, in consideration with spaceon lab table and dominant hand.

Easy setting anduseful Stand-base bath
Since the both shapes of the evaporator
stand-base and the water & oil bath havebeen improved to round, the bath can be
set front always in spite of any angles(positions) of the evaporator base. It ispossible to check bath temperature andenter temp. Setting without looking over.

Improvement for even easier operation, New functions to be implemented
Implementation of automatic
reverse revolution to be suited fordrying-out and concentration ofpowder and solid substance etc.
Direction of flask revolution (Clockwise orCounter-clockwise) can be set up. Even
automatic reversing is available. And, it (N1300) can be applied to dry-out of powderand dry-out & concentration of samples
including solid substance.

Addition of new anti-reverse coverthat protects against pool ofcondensed fluid

Anti-reverse coverProtection cover against fluid pool is addedat the foot of a condenser (Receiver flaskside). It protects fluid pool that happens toappear at sealing part when the condenseris tilted. Anti-reverse cover blocks entry ofcondensed fluid that flows inside ofglassware.

Possible to fix jack at any elevation
depending on flask’s shape due tonon-stepping positions

With exclusive option added,
capable to put and remove
insulation hose easily

The jack can be adjustable without definite
positons freely, that is different from theconventional evaporator. Since elevation
can be fixed in accordance with size andshape of sample flask at any positions, it iseasy to handle even when trap ball is used.

In use of optional one-touch connector and
one-touch insulation hose, the condenser andthe insulation hose can be put and removed
easily. One-touch insulation hose makes ziptie bundling unnecessary despite bundled
before. Line-connection is established just byinserting the insulation hoses into theconnectors which are on a condenser.

OptionOne-touch connector(ID10mm, including 2 pieces)
With "One-touch cold insulation hose set" used
together, hose can be connected and disconnectedeasily.

Cat. No. 267980

One-touch cold insulation hose set(-20 to 40°C as applicable temperature)
It is not necessary to bundle zip-tie around hose like theconventional connection. The connection is establishedjust by insert into the one-touch connector.
Dew preventive cover for rotary evaporator

Cold insulation hose set

Exclusive cover to reduce dew which appears on A condenser
and nozzle parts.

It is preventive against dew that appears whencooling media is circulated through, and reduces lossof cooling ability.

(-30 to 80°C as applicable temperature)

By putting exclusive cover on A condenser and nozzle parts, dew (dew condensation water)can be preventive. This cover can be applied to not only this new model but also models of
the conventional evaporators.

Product name

Dew preventive cover for rotary evaporator

One touch cooling hose set

Consist: Cover for condenser, Covers for nozzle parts including 2 sets
Material: PP, Insulation, Urethane foam
Using condition: Higher than -10˚C of circulating fluid temp.
(When circulating cooling media.)

Cooling hose set

Cat. No. 266040 for V type
Cat. No. 270730 for S type

Tube diameter
OD 10mm
ID 9mmLength

Cat. No.

2m244940

5m244950

2m1127005m174420

Relating products

Transparent cover to confirm evaporation status.

Please hook up the following EYELA products when A evaporator constitutes system.

Condenser cover for rotary evaporator

Low temp. circulator :CA-1115, CCA-1111
Diaphragm vacuum pump : NVP-1000, NPV-2000, NPV-2100
Solvent recovery unit :DPE series
Vacuum control unit :NVC-2300 seriesComposition: Condenser cover, Nozzle cover (2 pcs)
Material: Transparent PET, Silicone, Foamed silicone
Operating condition: Lowest circulating liquid temperature 5˚C(Room temp. 30˚C, Humidity less 70%, circulation liquid; water)

Cat. No. 266110

Polyurethane resin coating which preventsfrom glass scattering when broken.

We have also F series evaporator which includes the coated glassware (Eyela coat®) that issuperior with chemical resistance, transparency, cold resistance (-80°C), heat resistance (120°C).
Main unit composition

Glass parts that are applied with Eyela coat (Polyurethane
resin coating), are not broken easily while glass andsample are not scattered in pieces and splash easily even
if it is broken. Glass transparency is high. Membrane of
new composition which is superior for chemical property.
■ polyurethane resin that is not like vinyl chlorideof high environment load and is environment
freiendly, is selected.
■ Range of heat resistance temperature, is -80 to120°C. It is tough with each solvent and superior
with chemical resistance.

www.eyelaworld.com

Without bath
N-1300 only

With water bath
N-1300+SB-1300

With water/oil bath
N-1300+OSB-2200

Glass set
Eyela
COAT®

Eyela
COAT®

Eyela
COAT®

Model

AC230V
Cat. No.

AC115V
Cat. No.

EF

N-1300EF

266502266509VF

N-1300VF

266442266449SF

N-1300SF

266382266389EF

N-1300EF-W

266522266529VF

N-1300VF-W

266462266469SF

N-1300SF-W

266402266409EF

N-1300EF-WB

266542266549VF

N-1300VF-WB

266482266489SF

N-1300SF-WB

266422266429* Eyela coat® is applied to condenser, receiving flask, adaptor (V type).

Safety Caution

Please read "Instruction Manual" carefully before operation
for your safety.

TN Koishikawa Bldg.
1-15-17, Koishikawa, Bunkyo-ku,
Tokyo, 112-0002 Japan
TEL: +81-3-6757-3378
FAX: +81-3-3868-6571
E-mail: info.eyela@eyela.co.jp
Content of this catalogue is subject to change without prior noticeN-2016J5000-SA