VisionAire® 5

User Manual (EN)VisionAire 5
User Controls & System Status Indicators
ISO 7000Read user’s manual before operation.
Reg. # 1641
Keep away from rain, keep dry. Reg.
# 0626
Stacking limit by number. Reg. # 2403
Name and address of manufacturer.
Reg. # 3082
The country and date of manufacture.
The “CC” identifies the two lettercountry code of the country of manufacture. The date of manufacture is in
YYYY-MM-DD format. Reg. # 6049
Caution, consult accompanying documents. Reg. # 0434A
Catalog Number. Reg. # 2493

Warning. Reg. # W001
Council Directive 93/42/EEC
Authorized representative in theEuropean Community
If the product unique device identifier(UDI) label has the CE#### symbolon it, the device complies with therequirements of Directive 93/42/EEC
concerning medical devices. The
CE#### symbol indicates notified
body number.
Additional Symbols
Keep away from flammable materials,
oil and grease.
Do not disassemble.
When present on the device alarm
panel indicates external power interruption has been detected.

Serial Number. Reg. # 2498
Storage or operating temperature
limitation range Reg. # 0632

2018 Labeling: When present on thedevice alarm panel indicates low oxygen concentration in device output.

Storage humidity range
Reg. # 2620

ON (power switch on)

This way up. Reg. # 0623

OFF (power switch off)

Fragile, handle with care. Reg. #0621
Contains hazardous substances. Reg.
# 3723
Importer. Reg. # 3725
ISO 7010
The instruction manual must be read.
Reg. # M002
Keep away from open flame, fire,
sparks. Open ignition source andsmoking prohibited. Reg. # P003
Do not smoke near unit or while
operating unit. Reg. # P002
Type BF applied part (degree ofprotection against electric shock).
Reg. # 5333

2 - US PN MN238 E | User Manual

IEC 60417

Authorized representative in
Switzerland.
If the device bears the UKCA markas shown with UKCA#### indicating
the notified body number, this device
complies with UKCA regulations.
Class II equipment

21 CFR 801.15
Federal law restricts this device tosale by or on the order of a physician.
IEC 60601-1
Drip Proof Equipment - IP21

VisionAire 5
Council Directive 2012/19/EU
WEEE
This symbol is to remind the equipment owners to return it to a recycling
facility at the end of its life, per Waste
Electrical and Electronic Equipment(WEEE) Directive.
Our products will comply with therestriction of Hazardous Substances(RoHS) directive. They will not contain more than trace amounts of leador other hazardous material content.
ISO 15223-1
Medical Device. Reg. # 5.7.7
Unique device identifierReg. # 5.7.10
QPS Electrical Safety Certification
Safety agency for CAN/CSA
C22.2 No. 60601-1-14 for medical
electrical equipment. Certified forboth the U.S. and Canadian markets,
to the applicable U.S. and Canadianstandards.
This product may be covered by one or more patents, US and international. Please visit our website
below for the listing of applicable patents. Pat.:www.caireinc.com/corporate/patents/.

VisionAire Oxygen
Concentrator
This User Manual will acquaint you with CAIRE’s
VisionAire 5 Oxygen Concentrator (both 120 V and
230 V versions), and all variations available. Make
sure you read and understand all of the information contained in this guide before operating yourconcentrator. Should you have any questions, yourEquipment Provider will be happy to answer themfor you.

What is the Oxygen
Concentrator
The air we breathe contains approximately 21%
oxygen, 78% nitrogen, and 1% other gasses. In the
VisionAire oxygen concentrator, room air is drawn
into the machine through the air intakes. It thenpasses through an adsorbent material called molecular sieve. This material separates the oxygen fromthe nitrogen and allows only the oxygen to passthrough. The result is a flow of high-concentration
oxygen delivered to the user.
Note: There is never a danger of depleting the oxygenin a room when you use your oxygen concentrator
unit.

PN MN238 E | User Manual US - 3

VisionAire 5
Why Your Physician
Prescribed Oxygen
Many people suffer from a variety of heart, lung,and other respiratory diseases. A significant numberof these people can benefit from supplemental
oxygen therapy at home, when traveling, or while
participating in daily activities away from home.
Oxygen is a gas that makes up 21% of the room air
we breathe. Our bodies depend on a steady supply
to function properly. Your physician prescribed aflow or setting to address your particular respiratory
condition.
Although oxygen is a non-addictive drug, unauthorized oxygen therapy can be dangerous. You mustseek medical advice before you use this oxygen
concentrator. The Equipment Provider who supplies
your oxygen equipment will demonstrate how to set
the prescribed flow rate.
WARNING: “NO SMOKING – OXYGEN IN
USE” SIGNS MUST BE PROMINENTLYDISPLAYED IN THE HOME, OR WHEREOXYGEN IS IN USE. USERS AND THEIRCAREGIVERS MUST BE INFORMEDABOUT THE DANGERS OF SMOKING IN
THE PRESENCE OF, OR WHILE USING,
MEDICAL OXYGEN.
CAUTION: The Manufacturer recommends an alternate source of supplemental oxygen in the event of a power
outage, alarm condition, or mechanicalfailure. Consult your physician or Equipment Provider for the type of reserve
system required.
It is very important to select only the prescribed
level of oxygen. Do not change the flow selection
unless you have been directed to do so by alicensed clinician.
The Oxygen Concentrator may be used during
sleep under the recommendation of a licensed
clinician.

4 - US PN MN238 E | User Manual

Operator Profile
Concentrators are intended to supply supplemental
oxygen to users suffering from discomfort due toailments which effect the efficiency of one’s lungs
to transfer oxygen in the air to their bloodstream.
Stationary oxygen concentrators (SOCs) do notstore or contain oxygen. They do not need to be refilled, and can recharge anywhere AC or DC power.
Oxygen concentrator use requires a physician’sprescription and is not intended for life support use.
Although oxygen therapy can be prescribed forusers of all ages, the typical oxygen therapy user
is older than 65 years of age and suffers from avariety of respiratory diseases, including Chronic
Obstructive Pulmonary Disease (COPD). Users
typically have good cognitive abilities and must
be able to communicate discomfort. If the user isunable to communicate discomfort, or unable toread and understand the concentrator labeling andinstructions for use, then use is recommended only
under the supervision of one who can. If any discomfort is felt while using the concentrator, users
are advised to contact their healthcare provider.
Users are also advised to have back-up oxygen
available (i.e. cylinder oxygen) in the event of apower outage or concentrator failure. There areno other unique skills or user abilities required forconcentrator use.

VisionAire 5
Safety Features
The following information will acquaint you withsafety features of the VisionAire Oxygen Concentrator. Make sure you read and understand all theinformation contained in this manual before you
operate your unit. Should you have any questions,
your Equipment Provider will be happy to answer
them for you.

• Oxygen Monitor: The oxygen monitor detects
any drop in concentration below 82%. If this occurs the low oxygen concentration warning light() will illuminate. If the low O2 condition
persists, an audible intermittent alarm will also
activate in addition to the alarm light.
• Power Failure: In the event the unit is operating
and a loss of power occurs, the power warning
light (
) will illuminate and an audibleintermittent alarm will activate. See label image.
• Product Filter: ≥ 10 μm filter

Device warning label and alarm display.

• Compressor Motor: A pressure relief valve isfitted to the compressor outlet and is calibrated to
280 kPa (40 psig). Thermal safety is ensured bya thermostat situated in the stator winding of thecompressor (135°C / 275 °F).
• General Malfunction: If any of the conditions
listed below occurs, the general malfunction light) will illuminate and an audible intermittent(
alarm will activate.
This includes:• Obstruction to the flow of oxygen such as apinch or kink in the delivery cannula, triggered by high product tank pressure• High device product tank pressure conditionof greater than 33psig (±1)
• Low device product tank pressure conditionof less than 5psig (±1)• High device temperature of greater than
135°C (275 °F), triggered by low producttank pressure if the thermal switch locatedwithin the compressor trips (shutting down
the compressor)

PN MN238 E | User Manual US - 5

VisionAire 5
Unpacking Your VisionAire
Verify that all of the components listed are includedin the package. If any items are missing, contact
your oxygen provider immediately.
• Stationary Oxygen Concentrator

Getting to Know Your VisionAire Oxygen Concentrator
First, become familiar with the important parts ofyour VisionAire Oxygen Concentrator.
A. On/Off (I/0) Power Switch: Starts and stops theoperation of the unit.
B. Circuit Breaker Reset Button: Resets the unit
after electrical overload shutdown
C. Hour Meter: Records the unit’s total hours ofoperation.
D. Flowmeter/Adjustment Knob: Controls andindicates the oxygen flow rate in liters per minute(lpm).
E. Oxygen Outlet: Provides connection for ahumidifier (if required) or cannula.
F. Top and Side Handles: Enables convenience incarrying the unit.
G. Warning and Alarms Label
H. Specification Label: Displays electrical specifications and serial number.
I. Power Cord: Allows connection of unit into
electrical outlet.
J. Casters: Four casters enable unit to be easily
moved, as needed.
K. Humidifier Bottle Recess: Area to place theoptional humidifier bottle.
L. Humidifier Bottle (optional)
M. Humidifier Bottle Oxygen Outlet: Connection
for oxygen tubing/cannula.
N. Humidifier Bottle Oxygen Outlet
O. Humidifier Bottle Tubing
P. Humidifier Bottle Fitting
Q. Oxygen Tubing / Cannula

6 - US PN MN238 E | User Manual

WARNING: DO NOT USE EXTENSIONCORDS WITH THIS UNIT OR CONNECT
TOO MANY PLUGS INTO THE SAMEELECTRICAL OUTLET. THE USE OF
EXTENSION CORDS COULD ADVERSELY AFFECT THE PERFORMANCE OF
THE DEVICE. TOO MANY PLUGS INTOONE OUTLET CAN RESULT IN AN
OVERLOAD TO THE ELECTRICAL PANEL CAUSING THE BREAKER/FUSE TO
ACTIVATE OR FIRE IF THE BREAKEROR FUSE FAILS TO OPERATE.

VisionAire 5

F
E

C

L

F
B

D

K
A
G

G

J

Figure 1a

I
Q

N
H
O

P

Figure 1b

M

PN MN238 E | User Manual US - 7

VisionAire 5
Important!
Safety Instructions are defined as follows:
WARNING: IMPORTANT SAFETY INFORMATION FOR HAZARDS THAT MIGHT
CAUSE SERIOUS INJURY.
CAUTION: Important information forpreventing damage to the VisionAire
Family.
Note: Information needing special attention.

Indications for Use
Intended Use

The CAIRE VisionAire Oxygen Concentrator isintended for the administration of supplemental oxygen. The device is not intended for life support nor
does it provide any patient monitoring capabilities.
WARNING: FEDERAL (USA) LAW
RESTRICTS THIS DEVICE TO SALE OR
RENTAL BY ORDER OF A PHYSICIANOR OTHER LICENSED HEALTH CAREPROVIDER.

Contraindications for Use
WARNING: IN CERTAIN CIRCUMSTANCES, THE USE OF NON-PRESCRIBEDOXYGEN CAN BE HAZARDOUS. THIS
DEVICE SHOULD ONLY BE USED WHEN
PRESCRIBED BY A PHYSICIAN.
WARNING: NOT FOR USE IN THE PRESENCE OF
FLAMMABLE ANESTHETICS.
WARNING: AS WITH ANY ELECTRICALLYPOWERED DEVICE, THE USER MAY EXPERIENCEPERIODS OF NON-OPERATION AS A RESULT OF
ELECTRICAL POWER INTERRUPTION, OR THE
NEED TO HAVE THE OXYGEN CONCENTRATORSERVICED BY A QUALIFIED TECHNICIAN. THE
OXYGEN CONCENTRATOR IS NOT APPROPRIATE FOR ANY USER WHO WOULD EXPERIENCEADVERSE HEALTH CONSEQUENCES AS THE
RESULT OF SUCH TEMPORARY INTERRUPTION.

8 - US PN MN238 E | User Manual

WARNING: THIS UNIT IS NOT TO BE USED FOR
LIFE SUPPORT. GERIATRIC, PEDIATRIC, OR
ANY OTHER USER UNABLE TO COMMUNICATEDISCOMFORT WHILE USING THIS DEVICE MAY
REQUIRE ADDITIONAL MONITORING. USERSWITH HEARING AND/OR SIGHT IMPAIRMENT(S)
MAY NEED ASSISTANCE WITH MONITORING
ALARMS. IF YOU FEEL DISCOMFORT OR ARE
EXPERIENCING A MEDICAL EMERGENCY, SEEKMEDICAL ASSISTANCE IMMEDIATELY.

Safety Guidelines
WARNING: CAREFULLY REVIEW AND
FAMILIARIZE YOURSELF WITH THE
FOLLOWING IMPORTANT SAFETYINFORMATION ABOUT THE VISIONAIRE
INTENSITY OXYGEN CONCENTRATOR.
WARNING: IT IS VERY IMPORTANT TO SELECTONLY THE PRESCRIBED LEVEL OF OXYGEN. DO
NOT CHANGE THE FLOW SELECTION UNLESSYOU HAVE BEEN DIRECTED TO DO SO BY A
LICENSED CLINICIAN.
WARNING: DO NOT OPERATE THIS EQUIPMENTWITHOUT FIRST READING AND UNDERSTANDING
THIS MANUAL. IF YOU ARE UNABLE TO UNDERSTAND THE WARNINGS AND INSTRUCTIONS,CONTACT YOUR EQUIPMENT PROVIDER BEFOREATTEMPTING TO USE THIS EQUIPMENT; OTHERWISE INJURY OR DAMAGE COULD OCCUR.
WARNING: SMOKING WHILE USING OXYGEN IS
THE NUMBER ONE CAUSE OF FIRE INJURIES AND
RELATED DEATHS. YOU MUST FOLLOW THESE
SAFETY WARNINGS:WARNING: DO NOT ALLOW SMOKING, CANDLES,OR OPEN FLAMES IN THE SAME ROOM WITH THE
DEVICE OR THE OXYGEN-CARRYING ACCESSORIES.
WARNING: SMOKING WHILE WEARING AN OXYGEN CANNULA CAN CAUSE FACIAL BURNS AND
POSSIBLY RESULT IN DEATH.
WARNING: REMOVING THE CANNULA AND
PLACING IT ON CLOTHING, BEDDING, SOFAS,OR OTHER CUSHION MATERIAL WILL CAUSE A
FLASH FIRE WHEN EXPOSED TO A CIGARETTE,HEAT SOURCE, SPARK OR OPEN FLAME.

VisionAire 5
WARNING: IF YOU SMOKE, YOU MUSTALWAYS FOLLOW THESE THREE (3) IMPORTANT STEPS FIRST: TURN OFF THE
OXYGEN CONCENTRATOR, TAKE OFF
THE CANNULA, AND LEAVE THE ROOM
WHERE THIS DEVICE IS LOCATED.
WARNING: “NO SMOKING – OXYGEN IN USE”
SIGNS MUST BE PROMINENTLY DISPLAYEDIN THE HOME, OR WHERE OXYGEN IS IN USE.
USERS AND THEIR CAREGIVERS MUST BE INFORMED ABOUT THE DANGERS OF SMOKING IN
THE PRESENCE OF, OR WHILE USING, MEDICALOXYGEN.
WARNING: FEDERAL (USA) LAW RESTRICTS THIS
DEVICE TO SALE OR RENTAL BY ORDER OF APHYSICIAN OR OTHER LICENSED HEALTH CAREPROVIDER.
WARNING: THIS DEVICE SUPPLIES HIGH-CONCENTRATION OXYGEN THAT PROMOTES RAPID
BURNING. DO NOT ALLOW SMOKING OR OPEN
FLAMES WITHIN THE SAME ROOM OF (1) THIS
DEVICE, OR (2) ANY OXYGEN-CARRYING ACCESSORY. FAILURE TO OBSERVE THIS WARNING CANRESULT IN SEVERE FIRE, PROPERTY DAMAGE
AND / OR CAUSE PHYSICAL INJURY OR DEATH.
WARNING: DO NOT USE YOUR OXYGEN CONCENTRATOR IN THE PRESENCE OF FLAMMABLE
GASES. THIS CAN RESULT IN RAPID BURNINGCAUSING PROPERTY DAMAGE, BODILY INJURIESOR DEATH.
WARNING: DO NOT LEAVE A NASAL CANNULA
ON CLOTHING, BED COVERINGS OR CHAIRCUSHIONS. IF THE UNIT IS TURNED ON BUT NOT
IN USE, THE OXYGEN WILL MAKE THE MATERIAL FLAMMABLE. SET THE I/0 POWER SWITCH
TO THE 0 (OFF) POSITION WHEN THE OXYGENCONCENTRATOR IS NOT IN USE.
WARNING: USE NO OIL, GREASE, OR PETROLEUM-BASED OR OTHER FLAMMABLE PRODUCTS WITH THE OXYGEN-CARRYING ACCESSORIES OR THE OXYGEN CONCENTRATOR. OXYGENACCELERATES THE COMBUSTION OF FLAMMABLE SUBSTANCES. ONLY WATER BASED,OXYGEN COMPATIBLE LOTIONS OR SALVESSHOULD BE USED.
WARNING: DO NOT LUBRICATE FITTINGS, CONNECTIONS, TUBING, OR OTHER ACCESSORIESOF THE OXYGEN CONCENTRATOR TO AVOID THE
RISK OF FIRE AND BURNS.

WARNING: ELECTRICAL SHOCK HAZARD. TURN
OFF THE UNIT AND DISCONNECT THE POWER
CORD FROM THE ELECTRIC OUTLET BEFOREYOU CLEAN THE UNIT TO PREVENT ACCIDENTAL
ELECTRICAL SHOCK AND BURN HAZARD. ONLYYOUR EQUIPMENT PROVIDER OR A QUALIFIEDSERVICE TECHNICIAN SHOULD REMOVE THE
COVERS OR SERVICE THE UNIT.
WARNING: CARE SHOULD BE TAKEN TOPREVENT THE OXYGEN CONCENTRATORFROM GETTING WET OR ALLOWING FLUIDS TOENTER THE UNIT. THIS CAN CAUSE THE UNIT TO
MALFUNCTION OR SHUT DOWN, AND CAUSE AN
INCREASED RISK FOR ELECTRICAL SHOCK ORBURNS.
WARNING: DO NOT USE LIQUID DIRECTLY ON
THE UNIT. A LIST OF UNDESIRABLE CHEMICALAGENTS INCLUDES BUT IS NOT LIMITED TO THE
FOLLOWING: ALCOHOL AND ALCOHOL-BASEDPRODUCTS, CONCENTRATED CHLORINE-BASEDPRODUCTS (ETHYLENE CHLORIDE), AND OILBASED PRODUCTS (PINE-SOL®, LESTOIL®).
THESE ARE NOT TO BE USED TO CLEAN THE
PLASTIC HOUSING ON THE OXYGEN CONCENTRATOR, AS THEY CAN DAMAGE THE UNIT’S
PLASTIC.
WARNING: CLEAN THE CABINET, CONTROLPANEL, AND POWER CORD ONLY WITH A MILDDISINFECTANT APPLIED WITH A DAMP CLOTH(NOT WET) OR SPONGE, AND THEN WIPE ALL
SURFACES DRY. DO NOT ALLOW ANY LIQUID TOGET INSIDE THE DEVICE.
WARNING: THE OXYGEN CONCENTRATORSHOULD NOT BE USED ADJACENT TO OR
STACKED WITH OTHER EQUIPMENT. IF ADJACENT OR STACKED USE IS UNAVOIDABLE, THE
DEVICE SHOULD BE OBSERVED TO VERIFYNORMAL OPERATION.
WARNING: ALWAYS PLACE OXYGEN SUPPLYTUBING AND POWER CORDS IN A MANNER THATPREVENTS TRIP HAZARD OR POSSIBLE ACCIDENTAL STRANGULATION.
WARNING: NO MODIFICATION OF THIS EQUIPMENT IS PERMITTED.
WARNING: USE OF CABLES AND ADAPTERS
OTHER THAN THOSE SPECIFIED, WITH THE
EXCEPTION OF CABLES AND ADAPTERS SOLDBY THE MANUFACTURER OF THE MEDICAL ELECTRICAL EQUIPMENT AS REPLACEMENT PARTS
FOR INTERNAL COMPONENTS, MAY RESULT IN
INCREASED EMISSIONS OF DECREASED IMMUNITY OF THE OXYGEN CONCENTRATOR.
PN MN238 E | User Manual US - 9

VisionAire 5
WARNING: ENVIRONMENTAL CONDITIONS CAN AFFECT PERFORMANCE
OF DEVICE. LOCATE IN CLEAN, PESTFREE ENVIRONMENT.
WARNING: DEVICE SHOULD ONLY BE OPERATEDBY END USERS, TRAINED CARE-GIVERS OR
TRAINED TECHNICIANS. CHILDREN SHOULD NOTOPERATE DEVICE.
WARNING: TO ENSURE RECEIVING THE
THERAPEUTIC AMOUNT OF OXYGEN DELIVERYACCORDING TO YOUR MEDICAL CONDITION THE
VISIONAIRE MUST BE USED WITH THE SPECIFICCOMBINATION OF PARTS AND ACCESSORIESTHAT ARE IN LINE WITH THE SPECIFICATION OF
THE CONCENTRATOR MANUFACTURER AND
THAT WERE USED WHILE YOUR SETTINGS WERE
DETERMINED.
WARNING: USE OF THIS DEVICE AT AN ALTITUDE, TEMPERATURE, OR RELATIVE HUMIDITYOUTSIDE OF THE SPECIFIED VALUES LISTED IN
THIS MANUAL CAN ADVERSELY AFFECT THE
FLOWRATE AND THE OXYGEN CONCENTRATION,
AND CONSEQUENTLY THE QUALITY OF THE
THERAPY.
WARNING: THE USE OF SOME OXYGEN ADMINISTRATION ACCESSORIES NOT SPECIFIED FOR
USE WITH THIS OXYGEN CONCENTRATOR MAY
IMPAIR ITS PERFORMANCE. RECOMMENDEDACCESSORIES ARE REFERENCED WITHIN THISMANUAL.
WARNING: PREGNANT OR NURSING WOMENSHOULD NOT USE ACCESSORIES RECOMMENDED IN THIS MANUAL, THEY MY CONTAINPHTHALATES.

WARNING: USE ONLY ELECTRICAL VOLTAGE
AS SPECIFIED ON THE SPECIFICATION LABELAFFIXED TO THE DEVICE.
WARNING: DO NOT USE EXTENSION CORDS WITH
THIS UNIT OR CONNECT TOO MANY PLUGS INTO
THE SAME ELECTRICAL OUTLET. THE USE OF
EXTENSION CORDS COULD ADVERSELY AFFECT
THE PERFORMANCE OF THE DEVICE. TOO MANY
PLUGS INTO ONE OUTLET CAN RESULT IN AN
OVERLOAD TO THE ELECTRICAL PANEL CAUSING THE BREAKER/FUSE TO ACTIVATE OR FIRE IF
THE BREAKER OR FUSE FAILS TO OPERATE.
CAUTION: Federal (USA) law restricts
this device to sale or rental by order of aphysician or other licensed health careprovider.
CAUTION: Do not position the unit so that it isdifficult to access the power cord.
CAUTION: The concentrator should be located as
to avoid smoke, pollutants or fumes.
CAUTION: Ensure concentrator is operated in an
upright position.
CAUTION: Always place oxygen supply tubing
and power cords in a manner that prevents a trip
hazard.
CAUTION: Position the unit away from curtains ordrapes, hot air registers or heaters. Be certain toplace the unit on a flat surface and make sure allsides are at least 1 foot (30 cm) away from a wallor other obstruction. Do not place the unit in a confined area. Choose a dust and smoke free-location
away from direct sunlight. Do not operate the unit
outdoors unless the unit is plugged into a Ground
Fault Circuit Interrupter (GFCI) protected outlet.
CAUTION: Do not operate this unit in a restricted
or confined space where ventilation can be limited.
This can cause the device to overheat and affectperformance.
CAUTION: Do not allow either the air intake or the
air outlet vents to be blocked. DO NOT drop or
insert any object into any openings on the device.
This can cause the Oxygen Concentrator to overheat and impair performance.

10 - US PN MN238 E | User Manual

VisionAire 5
CAUTION: The Manufacturer recommends an alternate source of supplemental oxygen in the event of a power
outage, alarm condition, or mechanicalfailure. Consult your physician or Equipment Provider for the type of reserve
system required.
CAUTION: It is very important to select only theprescribed level of oxygen. Do not change the flow
selection unless you have been directed to do soby a licensed clinician.
CAUTION: The Oxygen Concentrator may be
used during sleep under the recommendation of alicensed clinician.
CAUTION: Operating or storing the Oxygen Concentrator outside of its normal operating temperature range can impair the performance of the unit.
Refer to the specification section of this manual forstorage and operating temperature limits.
CAUTION: In the event of an alarm or you observe
the Oxygen Concentrator is not working properly;
consult the troubleshooting section of this manual.
If you cannot resolve the problem, consult yourEquipment Provider.
CAUTION: If the audio alarm is weak or does notsound at all, consult your Equipment Provider
immediately.
CAUTION: If the humidifier bottle tubing is notproperly connected to the humidifier bottle fitting
or to the oxygen outlet, an oxygen leak can occur.
CAUTION: Normally, you should not need to adjust
the flowmeter on your unit. If you turn the flowmeter adjustment knob clockwise, you will decrease
and can shut off the flow of oxygen from your unit.
For your convenience, the flowmeter is marked in½ LPM increments. For units equipped with the 2
LPM flowmeter option, the flowmeter is marked in
1/8 LPM increments for flow settings up to 2 LPM.

Note: Cannula must be non-kinking, which can beused for a total length of 25 ft. (7.6 m) max.
Ensure the cannula is fully inserted and secure. You
should hear or feel oxygen flow to the prongs of thenasal cannula. If oxygen does not seem to flow, first
verify that the flow meter ball is registering a flow.
Then, place the tip of the cannula into a glass ofwater; if bubbles come out of the cannula, oxygen is
flowing. If bubbles do not appear, refer to the troubleshooting section of this manual.
Note: Always follow the cannula manufacturer’s
instructions for proper use. Replace the disposable
cannula as recommended by the cannula manufacturer or your Equipment Provider. Additional supplies
are available from your Equipment Provider.
Note: The VisionAire Oxygen Concentrator must be
operated for at least five minutes at 2 LPM before
using the unit.
The VisionAire is appropriate for usage by two users,
provided the combined flow is a minimum of 2 LPM
and does not exceed the maximum capacity of theconcentrator.
To Equipment Provider: The following oxygen administration accessories are recommended for use with
the VisionAire Oxygen Concentrator:• Humidifier Bottle: Part No. HU003-1• Nasal Cannula with 7 feet (2.1 m) of tubing (6 LPMmax): Part No. CU002-1
Note: The Manufacturer does not recommend thesterilization of this equipment.
Note: If the unit has not been used for an extended
period of time, it needs to operate for several minutes
before power failure alarm can become activated.
Note: The concentrator releases warm air out thebottom of the unit which can permanently discolor
temperature sensitive flooring surfaces such as vinyl.
The concentrator should not be used over flooring thatis sensitive to heat staining. The Manufacturer is not
responsible for flooring that becomes discolored.
Note: To prevent a void warranty, follow all manufacturers’ instructions.
Note: Do not attempt any maintenance other than thepossible solutions listed within the manual.
Note: Portable and mobile radio frequency (RF) communications equipment can effect medical electricalequipment.
Note: There is never a danger of depleting the oxygenin a room when you use your Oxygen Concentrator
unit.
PN MN238 E | User Manual US - 11

VisionAire 5
Note: To Equipment Provider: The following oxygen
administration accessories are recommended for use
with the VisionAire:• Nasal Cannula: CAIRE Part Number CU002-1• Humidifier Adaptor Tubing: CAIRE Part number
20843882• Humidifier Bottle: CAIRE Part Number HU003-1• Firebreak: CAIRE Part Number 20629671
A firebreak is recommended/required for use with any
cannula.
• CAIRE offers a firebreak intended to be used in conjunction with the oxygen concentrator. The firebreak is
a thermal fuse to stop the flow of gas in the event thatthe downstream cannula or oxygen tubing is ignited
and burns to the firebreak. It is placed in-line with thenasal cannula or oxygen tubing between the patient
and the oxygen outlet of the VisionAire. For properuse of the firebreak, always refer to the manufacturer’s instructions (included with each firebreak kit).
• Additional recommended accessories information isavailable online at www.caireinc.com.

12 - US PN MN238 E | User Manual

WARNING: KEEP OUT OF THE REACH
OF CHILDREN UNTIL INSTALLED.
WARNING: THIS PRODUCT CAN EXPOSE YOU
TO CHEMICALS INCLUDING NICKEL, WHICH
IS KNOWN TO THE STATE OF CALIFORNIA TO
CAUSE CANCER. FOR MORE INFORMATION, GO
TO WWW.P65WARNINGS.CA.GOV.
WARNING: IN THE EVENT THERE IS A SERIOUS
INCIDENT OCCURRING WITH THIS DEVICE, THE
USER SHOULD IMMEDIATELY REPORT THE
INCIDENT TO THE PROVIDER AND/OR THE MANUFACTURER. A SERIOUS INCIDENT IS DEFINEDAS AN INJURY, DEATH, OR POTENTIAL TO CAUSE
INJURY/DEATH SHOULD THERE BE A REOCCURRENCE OF THE INCIDENT. THE USER CAN ALSOREPORT THE INCIDENT TO THE COMPETENT AUTHORITY IN THE COUNTRY WHERE THE INCIDENTOCCURRED.

VisionAire 5

Specifications
VisionAire 5
Outlet Pressure

9 PSIG Max.

Flow Rates*

1 LPM – 5 LPM±10% of indicated setting, or 200 mL, whichever is greater*

Dimensions

14.1 in. W x 11.5 in. D x 20.8 in. H(35.8 cm W x 29.2 cm D x 52.8 cm H)

Weight30 lbs (13.6 kg)

Sound Pressure Level**

39,53 dB(A) ± 0,41 dB(A) @ 3 LPM

Power Consumption

290 Watts

O2 Concentration

90% (5.5% to / -3%)

Maximum Output Pressure

9 PSIG (62 KPa)

Electrical Requirements

115 VAC / 60 Hz, 3.0A
230VAC /50 Hz, 1.5A
230VAC / 60 Hz, 1.5A

Operating Environment*

5–40° C (41–104° F) at altitudes from up to 10,000 ft (3048 m) above sea
level. 15 - 90% relative humidity (non-condensing)

Altitude-1250 to 10,000 ft (-381–3048 m) (tested to 700–1060 hPa)

Storage Environment-25–70° C (-13–158°F). 0–90% relative humidity (non-condensing)* Based on an atmospheric pressure range of 700 hPa to 1060 hPa at 70°F (21°C)** Sound level measured per test method Nr. 14-1 10/2018 MDS-Hi.

The expected service life of the equipment is a minimum of five years.
See Technical Manual (PN MN138-1) for Sound Power Level.

WARNING: USE OF THIS DEVICE AT AN ALTITUDE, TEMPERATURE, OR RELATIVE HUMIDITYOUTSIDE OF THE SPECIFIED VALUES LISTED IN THIS MANUAL CAN ADVERSELY AFFECT THE
FLOWRATE AND THE OXYGEN CONCENTRATION, AND CONSEQUENTLY THE QUALITY OF THE
THERAPY.
WARNING: USE OF DEVICE OUTSIDE OF SPECIFIED OPERATING CONDITIONS IS EXPECTED TO ADVERSLY AFFECT THE FLOWRATE AND PERCENTAGE OF OXYGEN AND CONSEQUENTLY THE QUALITY OF THE
THERAPY.

PN MN238 E | User Manual US - 13

VisionAire 5
Operating Instructions
Review the following information before you operate your oxygen concentrator.
Note: The concentrator releases warm air out thebottom of the unit which can permanently discolor
temperature sensitive flooring surfaces such as vinyl.
The concentrator should not be used over flooring thatis sensitive to heat staining. The Manufacturer is not
responsible for flooring that becomes discolored.

Humidifier Bottle (Optional)

outlet, or to the optional humidifier’s oxygen outlet.
Note: Cannula must be non-kinking, which can beused for a total length of 50 ft. (15,2 m) max.
Ensure the cannula is fully inserted and secure. You
should hear or feel oxygen flow to the prongs of thenasal cannula. If oxygen does not seem to flow, first
verify that the flowmeter ball is registering a flow.
Then, place the tip of the cannula into a glass ofwater; if bubbles come out of the cannula, oxygen is
flowing. If bubbles do not appear, refer to the troubleshooting section of this manual.

If additional humidification is required with youroxygen therapy, perform the following steps eachtime you fill or clean the humidifier, which may
have been initially set up for your use.

Always follow the cannula manufacturer’s instructions
for proper use. Replace the disposable cannula asrecommended by the cannula manufacturer or yourEquipment Provider. Additional supplies are available
from your Equipment Provider.

1. Remove the humidifier bottle from the humidifier bottle recess.

Starting the Concentrator

2. Open the humidifier bottle. If you have a prefilled bottle, do not perform this step. Proceed tostep 5.
3. Fill the humidifier bottle with cool or cold water(distilled water is preferred) to the fill line indicated on the bottle. DO NOT OVERFILL.
4. Re-connect the top cover to the humidifier bottle.
5. Place the humidifier bottle in the humidifier
bottle recess on the back of the concentrator
and connect the humidifier bottle tubing to theoxygen outlet and the humidifier bottle fitting.
CAUTION: If the humidifier bottle tubing
is not properly connected to the humidifier bottle fitting or to the oxygen outlet,
an oxygen leak can occur.

Cannula Connection

Connect the tubing and cannula to the unit’s oxygen

14 - US PN MN238 E | User Manual

WARNING: “NO SMOKING – OXYGEN IN
USE” SIGNS MUST BE PROMINENTLYDISPLAYED IN THE HOME, OR WHEREOXYGEN IS IN USE. USERS AND THEIRCAREGIVERS MUST BE INFORMEDABOUT THE DANGERS OF SMOKING IN
THE PRESENCE OF, OR WHILE USING,
MEDICAL OXYGEN.
WARNING: DO NOT USE EXTENSION CORDS WITH
THIS UNIT OR CONNECT TOO MANY PLUGS INTO
THE SAME ELECTRICAL OUTLET. THE USE OF
EXTENSION CORDS COULD ADVERSELY AFFECT
THE PERFORMANCE OF THE DEVICE. TOO MANY
PLUGS INTO ONE OUTLET CAN RESULT IN AN
OVERLOAD TO THE ELECTRICAL PANEL CAUSING THE BREAKER/FUSE TO ACTIVATE OR FIRE IF
THE BREAKER OR FUSE FAILS TO OPERATE.
CAUTION: Position the unit away fromcurtains or drapes, hot air registers orheaters. Be certain to place the unit on
a flat surface and make sure all sides
are at least 1 foot (30 cm) away from awall or other obstruction. Do not place
the unit in a confined area. Choose adust and smoke free-location away fromdirect sunlight. Do not operate the unit
outdoors unless the unit is plugged intoa Ground Fault Circuit Interrupter (GFCI)
protected outlet.

VisionAire 5
CAUTION: Do not operate this unit in a restricted
or confined space where ventilation can be limited.
This can cause the device to overheat and affectperformance.
CAUTION: Do not allow either the air intake or the
air outlet vents to be blocked. DO NOT drop or
insert any object into any openings on the device.
This can cause the
CAUTION: Oxygen Concentrator to overheat andimpair performance.
1. Locate the unit near the electrical outlet in theroom where you spend most of your time.

Note: Optimal oxygen concentration is obtained within 10 minutes after the device is switched on (90%of the concentration is obtained after approximately
5 minutes).
Normally, you should not need to adjust theflowmeter on your unit. If you turn the flowmeter
adjustment knob clockwise, you will decrease andcan shut off the flow of oxygen from your unit.

Turning the Concentrator Off

Set the I/0 power switch to the “0” position to turnoff the unit.

2. Insert the power cord plug into the electricaloutlet.
3. Set the I/0 power switch to the “I” position toturn the unit on. An audible alarm will loudly
sound for approximately 1 seconds.
4. The low oxygen concentration indicator remains
on for a few minutes and until the oxygen
concentration reaches minimum concentration
requirements, (only pertains to unit equippedwith an Oxygen Monitor)
5. To set the flow of supplemental oxygen, turn theflowmeter adjustment knob left or right until theball inside the flowmeter centers on the flow line
number prescribed by your physician. To viewthe flowmeter at the proper angle, note that theback line and the front number line must give
appearance of one line.
6. The concentrator is now ready for use.
CAUTION: The Manufacturer recommends an alternate source of supplemental oxygen in the event of a power
outage, alarm condition, or mechanicalfailure. Consult your physician or Equipment Provider for the type of reserve
system required.
CAUTION: It is very important to select only theprescribed level of oxygen. Do not change the flow
selection unless you have been directed to do soby a licensed clinician.
CAUTION: The oxygen concentrator may be usedduring sleep under the recommendation of alicensed clinician
CAUTION: Ensure concentrator is operated in an
upright position.

PN MN238 E | User Manual US - 15

VisionAire 5
Cleaning, Care, and Proper MaintenanceCabinetWARNING: ELECTRICAL SHOCKHAZARD. TURN OFF THE UNIT AND
DISCONNECT THE POWER CORD
FROM THE ELECTRIC OUTLET BEFOREYOU CLEAN THE UNIT TO PREVENT
ACCIDENTAL ELECTRICAL SHOCK AND
BURN HAZARD. ONLY YOUR EQUIPMENT PROVIDER OR A QUALIFIED SERVICE TECHNICIAN SHOULD REMOVETHE COVERS OR SERVICE THE UNIT.
WARNING: CARE SHOULD BE TAKEN TOPREVENT THE OXYGEN CONCENTRATORFROM GETTING WET OR ALLOWING FLUIDS TOENTER THE UNIT. THIS CAN CAUSE THE UNIT TO
MALFUNCTION OR SHUT DOWN, AND CAUSE AN
INCREASED RISK FOR ELECTRICAL SHOCK ORBURNS.
WARNING: DO NOT USE OIL, GREASE, OR
PETROLEUM-BASED OR OTHER FLAMMABLE
PRODUCTS WITH THE OXYGEN-CARRYING ACCESSORIES OR THE OXYGEN CONCENTRATOR.
OXYGEN ACCELERATES THE COMBUSTION OF
FLAMMABLE SUBSTANCES.
WARNING: USE ONLY WATER-BASED LOTIONSOR SALVES THAT ARE OXYGEN COMPATIBLE PRIOR TO AND DURING THERAPY. NEVER USE PETROLEUM OR OIL-BASED LOTIONS OR SALVESTO AVOID THE RISK OF FIRE AND BURNS.
WARNING: DO NOT USE LIQUID DIRECTLY ON
THE UNIT. A LIST OF UNDESIRABLE CHEMICALAGENTS INCLUDES BUT IS NOT LIMITED TO THE
FOLLOWING: ALCOHOL AND ALCOHOL-BASEDPRODUCTS, CONCENTRATED CHLORINE-BASEDPRODUCTS (ETHYLENE CHLORIDE), AND
OIL-BASED PRODUCTS (PINE-SOL®, LESTOIL®).
THESE ARE NOT TO BE USED TO CLEAN THE
PLASTIC HOUSING ON THE OXYGEN CONCENTRATOR, AS THEY CAN DAMAGE THE UNIT’S
PLASTIC. CLEAN THE CABINET, CONTROL PANEL,
AND POWER CORD ONLY WITH A MILD DISINFECTANT APPLIED WITH A DAMP CLOTH (NOT WET)OR SPONGE, AND THEN WIPE ALL SURFACES
DRY. DO NOT ALLOW ANY LIQUID TO GET INSIDE
THE DEVICE.

16 - US PN MN238 E | User Manual

WARNING: CLEAN THE CABINET, CONTROLPANEL, AND POWER CORD ONLY WITH A MILDDISINFECTANT APPLIED WITH A DAMP (NOT
WET) CLOTH OR SPONGE, AND THEN WIPE ALL
SURFACES DRY. DO NOT ALLOW ANY LIQUID TOGET INSIDE THE CONCENTRATOR. PAY SPECIAL
ATTENTION TO THE OXYGEN OUTLET FOR THE
CANNULA CONNECTION TO MAKE SURE IT REMAINS FREE OF DUST, WATER, AND PARTICLES.
Note: Always follow the cannula manufacturer’s
instructions for proper use. Replace the disposable
cannula as recommended by the cannula manufacturer or your Equipment Provider. Additional supplies areavailable from your Equipment Provider.
Note: The Manufacturer does not recommend thesterilization of this equipment.
Use a mild disinfectant applied with a damp clothor sponge to clean the exterior of the concentrator,
and then wipe all surfaces dry. Do not allow any
liquid to get inside the device. Device cabinet
should be cleaned at a minimum between users.

Humidifier Bottle (optional)• Check water level daily and add water as needed• To clean and disinfect the humidifier, follow yourEquipment Provider’s instructions, or the instructions included with the humidifier bottle.

VisionAire 5
Alarm Conditions
All alarms are low priority alarms.
Alarm

Indicates

ActionGeneral malfunction

high product tank pressure

yellow lightand intermittent
audible alarm

OR

Ensure flowmeter is open to minimum flow rate or higher.
Ensure cannula is not kinked or obstructed. Remove any
devices connected downstream of the outlet of the device.
Ensure device has at least 12" of clearance on all sides
and intakes are not obstructed. Ensure external grossparticle intake filter is clean and not clogged. Ensure unit
is within operating temperature range. If issue persists,
contact equipment provider for service.

low product tank pressureORhigh device temperature

Oxygen monitor yel-

low oxygen concentration

Contact equipment provider for service.

power failure

Ensure device is plugged into a known, working outlet. Ensure breaker switch is pushed in. If issue persists contact
equipment provider for service.

low light
andintermittent audible
alarm
Power failure yellow
light
and
intermittent audiblealarm

PN MN238 E | User Manual US - 17

VisionAire 5
Troubleshooting
If your VisionAire Oxygen Concentrator fails to operate properly, refer to the chart on the following pages
for possible causes and solutions and, if needed, consult your Equipment Provider.
If you cannot get the unit to operate, connect your nasal cannula, face mask, or other accessories to areserve supplemental oxygen supply.
Note: Do not attempt any maintenance other than the possible solutions listed within this manual. Maintenance isthe responsibility of the provider and will be tracked by the provider.
Note: To prevent a void warranty, follow all manufacturers’ instructions.
Note: If the unit has not been used for an extended time period, it needs to operate for a minimum of 15 minutes
before power failure alarm can become activated.
Problem

Probable Cause

Solution

Unit does not operate. Powerfailure condition causes analarm to sound.

Power cord not connected
into electrical outlet.

Check power cord plug at the electrical outlet fora proper connection.

No power at electrical outlet.

Check power source, wall switch, fuse, or circuit
breaker in-house.

Oxygen concentrator circuit
breaker is activated.

Press (do not hold in) the circuit breaker reset
button in the front of the unit.
If the circuit breaker trips again or the alarm
continues to sound after the unit is turned on,contact your Equipment Provider.

Limited oxygen flow.

Condensation collects in theoxygen tubing when you use
the humidifier bottle.

Dirty or obstructed humidifier
bottle.

Remove the humidifier bottle from the oxygen
outlet. If flow is restored, clean or replace with a
new humidifier bottle.

Defective nasal cannula,face mask, catheter, and/or
oxygen delivery tube, or otheraccessory.

Remove and check accessories for kinks or
obstructions. Replace if needed.

Cannula tubing loose.

Check cannula tubing connection at control
panel.

Unit not properly ventilated.
Elevated operating temperature.

Make sure unit is positioned away from curtains
or drapes, hot air registers, heaters, and fireplaces. Be certain to place the unit so all sides areat least 12 inches (30.5 cm) away from a wallor other obstruction. Do not place the unit in aconfined area.
Refill humidifier bottle with COLD water.
DO NOT OVERFILL. Allow oxygen tubing to dry
out, or replace with new tubing.

Intermittent alarm sounds.

18 - US PN MN238 E | User ManualSee ‘Safety Features’ section
for a description of auditory
indicators.

Set I/0 power switch to 0 position, use yourreserve oxygen supply and consult your Equipment Provider immediately.

VisionAire 5
Problem

Probable Cause

Solution

Oxygen monitor light remainslit (yellow).

Oxygen concentration is≤ 82%.

Set I/0 power switch to the 0 position, use yourreserve oxygen supply (if provided), and consult
your Equipment Provider immediately.

All other problems.

Set I/0 power switch to the 0 position, use yourreserve oxygen supply and consult your Equipment Provider immediately.

Accessories
For proper performance and safety, use only theselisted accessories supplied by CAIRE throughyour oxygen provider. Use of accessories not listed
below could adversely affect the performance and/or safety of the concentrator.
VisionAire Family Standard Accessories
Humidifier Bottle(6-15 LPM)

Part Number – HU003-1

Cannula, 25 feet
(7.6 m) (6 LPM max)Part Number – CU002-4

Humidifier Bottle
TubingPart Number – TU255-1

Humidifier Bottle
Fitting

Part Number – F0655-1

Oxygen Tubing, 25
feetPart Number – CU004-3

Tubing/Cannula
Connector

Part Number – CU009-1

Note: Additional options may be available for country-specific power cords where noted above. Contact
CAIRE or your oxygen provider if alternate options areneeded for order.

PN MN238 E | User Manual US - 19

VisionAire 5
EMC Testing
Medical equipment needs special precautions regarding EMC and needs to be installed and put into service according to the EMC information provided in this section.

Guidance and Manufacturer’s Declaration­—Electromagnetic Emissions
The VisionAire is intended for use in the electromagnetic environment specified below. The customer or the user ofthe VisionAire should assure that it is used in such an environment.
Emissions Test
RF emissions
CISPR 11
RF emissions
CISPR 11
Harmonic emissions
IEC 61000-3-2
Voltage fluctuations/
flicker emissions
IEC 61000-3-3ComplianceElectromagnetic Environment - GuidanceGroup 1The VisionAire uses RF energy only for its internal function. Therefore, its RF emissions are very low and are not likely to cause any
interference in nearby electronic equipment.

Class B
Complies
Complies

The VisionAire is suitable for use in all establishments, including
domestic establishments and those directly connected to the
public low-voltage power supply network that supplies buildings
used for domestic purposes.

Recommended separation distances between portable and mobile RFcommunications equipment and the VisionAire Units

The VisionAire is intended for use in an electromagnetic environment in which radiated RF disturbances are controlled. The customer or the user of the VisionAire can help prevent electromagnetic interference by maintaining aminimum distance between portable and mobile RF communications equipment (transmitters) and the VisionAire asrecommended below, according to the maximum output power of the communications equipment.
Rated maximum
Separation distance according to frequency of transmitter
output power ofmtransmitterfrom 150 kHz to 80 MHz
from 80 MHz to 800 MHz
from 800 MHz to 2,5 GHzd  1.2 P
d  1.2 P
d  1.2 P
W
0.01
0.12
0.12
0.23
0.1
0.38
0.38
0.73
1
1.2
1.2
2.3
10
3.8
3.8
7.3
100
12
12
23
For transmitters rated at a maximum output power not listed above, the recommended separation distance d in metres (m) can be estimated using the equation applicable to the frequency of the transmitter, where P is the maximum
output power rating of the transmitter in watts (W) according to the transmitter manufacturer.
NOTE 1 At 80 MHz and 800 MHz, the separation distance for the higher frequency range applies.
NOTE 2 These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorption andreflection from structures, objects and people.

20 - US PN MN238 E | User Manual

VisionAire 5
Guidance and Manufacturer’s Declaration—Electromagnetic Immunity

The VisionAire is intended for use in the electromagnetic environment specified below. The customer or the user ofthe VisionAire should assure that it is used in such an environment.
Immunity test
IEC 60601 test level
Compliance level
Electromagnetic environment – guidanceElectromagnetic
Floors should be wood, concrete orenvironment – guid- ±6 kV contact±6 kV contact
ceramic tile. If floors are covered withance
synthetic material, the relative humidity±8 kV air
±8 kV airshould be at least 30 %.
IEC 61000-4-2
Electrical fasttransient/burst
IEC 61000-4-4
Surge
IEC 61000-4-5±2 kV for power supplylines±1 kV for input/output
lines±1 kV line(s) to line(s)±2 kV line(s) to earth<5 % UT(>95 % dip in UT )for 0,5 cycle
40 % UT

Voltage dips, shortinterruptions andvoltage variations
on power supply
input lines IEC
61000-4-11(60 % dip in UT )for 5 cycles
70 % UT(30 % dip in UT )25 cycles<5 % UT

N/A

Mains power quality should be that of atypical commercial or hospitalenvironment.

±1 kV line(s) to line(s) Mains power quality should be that of atypical commercial or hospital±2 kV line(s) to earth environment.
<5 % UT(>95 % dip in UT )for 0,5 cycle
40 % UT(60 % dip in UT )for 5 cycles
70 % UT(30 % dip in UT )for 25 cycles

Mains power quality should be that of atypical commercial or hospital environment. If the user of the VisionAire requires continued operation during power
mains interruptions, it is recommended
that the VisionAire is powered from anuninterruptible power supply (UPS).

<5 % UT(>95 % dip in UT )(>95 % dip in UT )

3A/m

3A/m

for 5 sec

Power frequency
magnetic field IEC
61000-4-8±2 kV for power
supply lines

for 5 sec

Power frequency magnetic fields should
be at levels characteristic of a typical location in a typical commercial or hospitalenvironment.

NOTE UT is the A.C. mains voltage prior to application of the test level.

PN MN238 E | User Manual US - 21

VisionAire 5
Guidance and Manufacturer’s Declaration–Electromagnetic Immunity

The VisionAire is intended for use in the electromagnetic environment specified below. The customer or the user ofthe VisionAire should assure that it is used in such an environment.
Immunity test
IEC 60601 test level
Compliance level
Electromagnetic environment –Conducted RF

3 VrmsIEC 61000-4-6

150 kHz to 80 MHz

Radiated RF

3 V/m

IEC 61000-4-3

80 MHz to 2,5 GHz

3 Vrms3 V/mguidance
Portable and mobile RF communications
equipment should be used no closer
to any part of the VisionAire, including
cables, than the recommended separation distance calculated from theequation applicable to the frequency of
the transmitter.
Recommended separation distanced  1.2 P
d  1.2 P = 80 MHz to 800 MHz
d  1.2 P = 800 MHz to 2,5 GHz
where P is the maximum output power
rating of the transmitter in watts (W)according to the transmitter manufacturer
and d is the recommended separation
distance in metres (m).
Field strengths from fixed RF transmitters, as determined by an electromagnetic site survey,a should be less thanthe compliance level in each frequency
range. b
Interference may occur in the vicinityof equipment marked with the following
symbol:NOTE 1 At 80 MHz and 800 MHz, the higher frequency range applies.
NOTE 2 These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorption andreflection from structures, objects and people.
aField strengths from fixed transmitters, such as base stations for radio (cellular/cordless) telephones and land
mobile radios, amateur radio, AM and FM radio broadcast and TV broadcast cannot be predicted theoretically withaccuracy. To assess the electromagnetic environment due to fixed RF transmitters, an electromagnetic site survey
should be considered. If the measured field strength in the location in which the VisionAire is used exceeds theapplicable RF compliance level above, the VisionAire should be observed to verify normal operation. If abnormal
performance is observed, additional measures may be necessary, such as re-orienting or relocating the VisionAire.
b

Over the frequency range 150 kHz to 80 MHz, field strengths should be less than 3 V/m.

22 - US PN MN238 E | User Manual

VisionAire 5
Method of Disposal

Classification

Disposal of Waste

Type of protection against electric shock:Class II
Protection from electric shock is			achieved by double insulation.
			Protective earthing or reliance upon			
installation conditions are not			required.

All waste from CAIRE’s VisionAire Oxygen Concentrator must be disposed using the appropriate
methods specified by local authorities

Disposal of Device

In order to preserve the environment, the concentrator must be disposed using the appropriate methods specified by local authorities.

Degree of protection against electric shock:Type BF Equipment providing a particular		 degree of protection against electric
		 shock regarding		 1) allowable leakage current;		 2) reliability of protective earth connection
		 (if present).
		 Not intended for direct cardiac application.
Degree of protection against harmful ingress ofwater:
		 Drip-proof equipment – IP21
		 Protection against ingress of solid foreign
		 objects greater than 12.5 mm diameter.
		 Equipment provided with an enclosure		 preventing of such an amount of falling liquid		 as might interfere with the satisfactory and		 safe operation of the equipment.
Method of cleaning and infection control allowed:
		 Please refer to Maintenance section in the		 VisionAire Service Manuals.
Degree of safety of application in the presence offlammable anesthetic gases:		 Equipment not suited for such application.
Mode of operation:
Continuous duty.
Note: Always follow the cannula manufacturer’s
instructions for proper use. Replace the disposable
cannula as recommended by the cannula manufacturer or your Equipment Provider. Additional supplies areavailable from your Equipment Provider.
Note: The manufacturer does not recommend thesterilization of this equipment.

PN MN238 E | User Manual US - 23

www.c a ire inc .c om

CAIRE Inc.
2200 Airport Industrial Dr., Ste. 500
Ball Ground, GA 30107 U.S.A.
Medical Product Service GmbH
Borngasse 20
35619 Braunfels, Germany
Tel: +49 (0) 6442-962073
E-mail: info@mps-gmbh.eu
Accumed Sagl
Viale Serfontana, 106834 - Morbio Inferiore
Switzerland
CAIRE and CAIRE Inc. are registered trademarks of CAIRE Inc. Please
visit our website below for a full listing of trademarks. Trademarks: www.
caireinc.com/corporate/trademarks/.
Copyright © 2022 CAIRE Inc. CAIRE Inc. reserves the right todiscontinue its products, or change the prices, materials, equipment,
quality, descriptions, specifications and/or, processes to its products
at any time without prior notice and with no further obligation orconsequence. All rights not expressly stated herein are reserved by us,
as applicable.

|MN238-E$~

21 Sept 2022 PN MN238 E