Operating Instruction

Centrifuge LC 24

Instruction ManualCentrifuge LC 24
with horizontal rotor 6 x
SARSTEDT No.: 90.184.700

V. 3.1 / 07.2006

Operating Instruction

Centrifuge LC 24

Table of contents
page E-

1. Device data ........................................................................................................................ 3
2. Manufacturer's address ...................................................................................................... 3
3. Important notes................................................................................................................... 3
Explanation of symbols.................................................................................................................4
Device-specific safety notes .........................................................................................................4
Responsibilities of the owner........................................................................................................5
Responsibilities of operating personnel ........................................................................................5

4. Other information
Guarantee and liability..................................................................................................................6
Shutdown / Disposal.....................................................................................................................6

5.Description of the centrifuge................................................................................................ 7
6. Transport, Installation and Connection ............................................................................... 7
Delivery Schedule ........................................................................................................................7
Unpacking ....................................................................................................................................8
Technical Data .............................................................................................................................8
Installation....................................................................................................................................8
Commencing Operation................................................................................................................9

7. Operation.......................................................................................................................... 10
Control Elements........................................................................................................................10
Functions and Settings...............................................................................................................11
Time Setting and Continuous Running ............................................................................11
Speed / RCF....................................................................................................................11
Acceleration Rate ............................................................................................................11
Braking Rate....................................................................................................................12Countdown Delay ............................................................................................................12
Sensitivity of the Imbalance Trip ......................................................................................12
Audible Signals................................................................................................................13
Saving and Retrieving Settings........................................................................................13
Counter ......................................................................................................................................14
Access to the Rotor Chamber during a Power Failure ................................................................14
Loading the Rotor.......................................................................................................................14
Centrifuging................................................................................................................................15

8. Maintenance and servicing
Service address .........................................................................................................................16
Cleaning.....................................................................................................................................16
Regular Maintenance .................................................................................................................17
Index ..........................................................................................................................................17
Contamination questionnaire......................................................................................................18

page E-2

V. 3.1 / 07.2006

Operating Instruction

Centrifuge LC 24

1. Device data
(to be completed by customer)
Type:______________________________________________________________________________________________________

Serial no.:___________________________________________________

Place of installation:___________________________________________________

Date of installation:___________________________________________________

Inventory no.:___________________________________________________

2. Manufacturer's address
SARSTEDT AG & Co.
Postfach 1220
D-51582 Nümbrecht
Phone: +49 (0) 22 93 / 305 0
Fax:+49 (0) 22 93 / 305 282
E-Mail: info@sarstedt.com

3. Important notes
Read the safety instructions in this manual
before putting the device into operation!
A sound knowledge of the contents of these operating instructions is a basic
requirement to ensure correct use and trouble-free operation of the device.
Please keep these instructions in a safe place for future reference.
Copyright
The copyright for these operating instructions lies with SARSTEDT-AG & Co. The instructions are
only intended for use by operating personnel and for the purchaser of the device.
Reproduction or distribution of this publication, in whole or in part, is not permitted without the writtenconsent of SARSTEDT AG & Co.
All rights reserved. Infringements can be prosecuted.
Sarmix and Sedivette are registered trade names of SARSTEDT AG & Co.
Technical modification reserved.
SARSTEDT AG & Co.

V. 3.1 / 07.2006

page E-3

Operating Instruction

Centrifuge LC 24

Explanation of symbols
These operating instructions contain important symbols to indicate dangers and possible operating
errors.
The ”Caution“ symbol means that in this section:• Important information is given.
• An immediate or possible risk of danger to life and limb of persons is
described.
• A note on dangerous situations is given.
Non-observance of these symbols can lead to minor, serious, or evenlife-threatening injuries as well as to material damage.

Warning of hazardous electrical voltage!

Warning against harmful or irritating substances!
• Failure to observe this symbol can result in severe harm to health.
• Always comply with the applicable accident prevention regulations (APR)
when handling these substances.
This “Info” symbol refers to notes regarding proper use of the device or to general,
explanatory information. Non-observance of these notes or information can lead to thedevice being operated incorrectly or even becoming damaged.
Additional symbols contain notes on the topic displayed in the symbol, for example
accident prevention regulations, supply voltage, maintenance, or disposal.

Device-specific safety notes
This device has been built according to the latest state of the art and the accepted
principles of safety engineering.
All relevant safety information and guidelines, as well as regulations for safety at workand accident prevention regulations for laboratory use must be observed.
Nonetheless, danger can arise for the user or third parties during operation, as well as
damage to the device or other material assets.
The device may only be used:• for the intended purpose• in technically safe and perfect condition.
• in adherence with the current accident prevention regulations.
• for the separation of materials of varying density or particle size suspended ina fluid.
• with a maximum sample density of 1.15 g/cm3.
The device may not be used:• in explosion-hazarded areas.
• with flammable or explosive materials or substances that may precipitatestrong reactions with one another.
• for centrifugation of toxic or radioactive materials and pathogenic microorganisms without appropriate safety systems.

page E-4

V. 3.1 / 07.2006

Operating Instruction

Centrifuge LC 24

During the warranty period, the device may only be repaired by SARSTEDT AG & Co.
or by persons authorized by SARSTEDT AG & Co.
In case of improper use or repair, the warranty will be void.
The device is a piece of laboratory equipment and complies with the international
standard EN 61010 (safety requirements for laboratory equipment).
The device is operated with a mains voltage of 230 V/50-60 Hz. Therefore, thefollowing notes must be observed:• No water or other liquids may be permitted to enter the device.
• The device may only be connected to the mains supply by means of a suitable
power cord supplied.
• Before operating the device, check the power cord for signs of damage.
• Damaged power cords must not be used under any circumstances.
• The device must only be connected to the mains voltage specified on the typelabel.
• A damaged device must not be operated.
• Never open housing parts that are fixed with screws.
• Remove the plug from the wall outlet, if the device is not used for a prolonged
period or time if it is to be cleaned.
The device must only be used with the accessories described in these operating
instructions.

Responsibilities of the owner
The device owner is responsible for ensuring that only such persons are allowed to operate thedevice, who have read these instructions and understand them. This should be confirmed by the
person's signature.
Moreover, the observance of safety regulations by personnel should be checked at regular intervals.

Responsibilities of operating personnel
Before working with the device for the first time, the operator must:• read these operating instructions.
• observe the applicable rules and the regulations for accident prevention.

V. 3.1 / 07.2006

page E-5

Operating Instruction

Centrifuge LC 24

4. Other information
Guarantee and liability
The “Delivery and Payment Conditions” of SARSTEDT AG & Co. always apply. These are printed on
the reverse side of the invoice.
Guarantee and liability claims are excluded, if they are based on one or more of the following causes:•
use of the device for other than the intended purpose.
•
improper installation, commissioning, operation, or maintenance of the device.
•
operation of the device with defective safety fittings or incorrectly attached or non-functionalsafety or protective equipment.
•
non-observance of the notes in these operating instructions regarding transport, storage,
installation, commissioning, operation, maintenance, setting up, or disposal.
•
unauthorized modifications to the device.
•catastrophic failure due to external cause and/or force majeure.
•incorrect repairs.

Shutdown / Disposal••

If the device is to be scrapped, it must be handled and disposed of in accordancewith the applicable laws and regulations.
All substances used in connection with the device must be handled and disposedof in accordance with the applicable laws and regulations.

These operating instructions consist of the following materials:
Back binder and protective cover are of PVC – the rest is paper.
Remove the paper by tearing off the back binder.

page E-6

V. 3.1 / 07.2006

Operating Instruction

Centrifuge LC 24

5. Description of the centrifuge
The centrifuge LC 24 is a table-top centrifuge with horizontal rotor designed for continuous operation.
The horizontal rotor included in the standard supply schedule along with the tube holders can accept
up to 24 sample tubes. With the LC 24, almost every operating parameter from the acceleration andbraking rate to the time and speed settings can be changed. The high speeds and centrifugal forces
produce plasma low in thrombocytes to plasma entirely free of thrombocytes, and reduce processing
times to a very great extent. The special horizontal rotor allows fast and simple loading of samples
along with completely horizontal separation. You can save your settings in one of the 10 memory
locations and call them up at the touch of a button
A cover safety switch prevents the centrifuge from operating when the cover is open. The yellow“LOCK” display at the front of the centrifuge lights up as soon as the cover has been correctly locked.
In addition to the cover safety switch, the LC 24 is equipped with a “0 rpm” cover locking system.
This safety system keeps the cover closed at all times (even in the event of a power failure) andensures that the cover can be unlocked only when the rotor has come to a standstill.

6. Transport, Installation and Connection
Delivery ScheduleStandard delivery schedule
1
6
6
1
1

x
x
x
x
x

Centrifuge LC 24 horizontal rotor 6 x, SARSTEDT No. 90.184.700
Tube holder, 4x 100 mm
Caps for tube holders
Power cord
Operating instructions

Accessories
92.184.751
92.184.752
92.184.753
92.184.754
92.184.755

Tube holder, 4x for 100 ml
Tube holder, 1x for 50 ml
1.0 inch rubber adapter
0.25 inch rubber adapter
Caps for tube holders(recommended for glass and flat bottom tubes)(recommended for glass and flat bottom tubes)

If you have any questions concerning other SARSTEDT products or the SARSTEDT product range,
please contact:
SARSTEDT AG & Co.
Postfach 1220
D-51582 Nümbrecht
Phone:
Fax:+49 (0) 22 93 / 305 0+49 (0) 22 93 / 305 282

or contact your local SARSTEDT sales office.

V. 3.1 / 07.2006

page E-7

Operating Instruction

Centrifuge LC 24

Unpacking••

Check that the packaging shows no signs of damage.
In case of transport damage, please contact the shipping agent immediately.
Observe the notification periods of the different freight carriers (railway, mail,parcel service, trucking company, etc.). It can be as short as 24 hours.
Check the scope of delivery for completeness.
Defects and damage must be notified immediately to SARSTEDT AG & Co.

••
Technical Data
Machine data
Machine:Order No.:

Centrifuge LC 24
SARSTEDT No. 90.184.700

Centrifuge motor:
Timer:

HP brushless induction motor
electronic,
continuous operation or 1-99 minutes +/- 1%
max. 3,500 rpm
max. 2,500 x g

Speed:
RCF:
SupplyElectrical connection:
Power consumption:
Fuses:

0230 V ∼, ±10 %, 50-60 Hz
200 Watt
4.0 A, can be reset

Ambient conditions
Permissible ambient temperature:
Maximum relative humidity:+2 °C to +40 °C
90 %, non-condensing

Dimensions
Width x Depth x Height:
Weight:370 x 430 x 230 mm
approx. 18 kg

Rotor loading
Max. tube size
17 mm x 100 mm
17 mm x 125 mm
34 mm x 115 mm

Number of tubes
24
12
6

Installation
•
•
•
•
•
•
•
•

Position the centrifuge on a firm, vibration-free and flat surface.
Maintain a safety zone around the centrifuge of at least 30 cm.
During centrifuging, no person or dangerous substance must come within thesafety zone.
During operation, no contact must occur with other equipment or objects.
Do not place vibration-sensitive objects or samples on the installation surface.
There must be sufficient space above the installation surface to open thecentrifuge cover.
Ensure that there is adequate air circulation to prevent the samples and thecentrifuge from overheating.
The installation location must always be well ventilated.

page E-8

V. 3.1 / 07.2006

Operating Instruction

Centrifuge LC 24

Commencing Operation
Caution before first use!
• Check that the actual supply voltage corresponds with the stated supply voltage.
• Comply with the permitted ambient conditions for the machine.
• In the cold season, after delivery and unpacking, wait approx. 1 hour until thecentrifuge has reached ambient temperature (18 °C to 23 °C).
• The control elements are shown in Section 7.
For safety reasons, the locking system is always activated.
• The user can switch it off for approx. 15 seconds by briefly pressing the“OPEN / STOP” button. This is possible only with the centrifuge switched on
and the rotor at a standstill.
• At the end of a centrifugation, the cover is automatically unlocked for approx.
60 seconds.
In order to exclude transport damage and to check correct installation, carry out a test run whenstarting up the centrifuge.
Procedure:• Connect the centrifuge to the power supply with the power cord.
• Switch on the centrifuge at the main switch.
• The main switch lights up.
• Press “OPEN / STOP” button.
• “UNLOCKED” display lights up.
• Turn the cover knob counter-clockwise and open the cover.
• There must be no foreign bodies in the rotor chamber.
• Turn the rotor by hand checking that it is running freely and horizontally, and that thescrew in the middle of the rotor is tight. If this is not the case, do not start up thecentrifuge!
• Insert the tube holders without any load and check that they are properly seated.
• The rotor must be evenly loaded.
• Close the cover and turn the cover knob fully clockwise.
• “LOCK” display lights up.
• The centrifuge will start running only when the cover is properly locked!
• Select the maximum speed of 3,500 rpm with the “Arrow buttons [A]”.
• Press the “START” button and carry out a test run.
• The “RUNNING” display lights up.
• The centrifuge accelerates to the correct speed. The tube holders slide into ahorizontal position.
• Check the centrifuge for running noises! A soft humming noise should be heard. In theevent of loud or unusual noises, stop the centrifuge immediately by pressing the“OPEN/STOP” button and then check the previous points once more.
• Press the “OPEN/STOP” button.
• The test run comes to an end and the rotor is braked to a standstill.
• The centrifuge is now ready to operate.

V. 3.1 / 07.2006

page E-9

Operating Instruction

Centrifuge LC 24

7. Operation
Control Elements

2
1
3

4

5

6

1
4

Cover
Control panel

2
5

7

Cover knob
Counter83
6

Safety lock
Main switch

11

10

9

13
12

20

14

19

7
10
13
16
19

Display [A]“LOCK” (cover closed)“STOP/OPEN” button“PROGRAM” button
Display [B]

18

8
11
14
17
20

16

15

Arrow buttons [A]“UNLOCKED”“MEMORY” button
Braking rate
Switching button RCF

9
12
15
18

page E-10

17“Running” (operating)“START” button
Memory location
Arrow buttons [B]V. 3.1 / 07.2006

Operating Instruction

Centrifuge LC 24

Time Setting and Continuous Running••

The running time can be set for periods of between 00:30 minutes and 99:30
minutes.
Changes to settings can also be made during a run.

Time setting:Enter settings by using the arrow buttons [B]. The settings are then shown in the display [B]. Whenthe centrifuge is started with this setting, the remaining running time is shown in the display [B].
Pressing the arrow buttons once changes the set time by 30 seconds. When a button is kept
depressed, each change is increased to a 1-minute step, while continuous pressure raises this to a 5minute step.
Continuous running:
Press the arrow buttons [B] until the display [B] shows “HOLD”. When the centrifuge is started withthis setting, the running time appears in the display [B]. Continuous running is automatically ended
after 99:30 minutes or before that by pressing the “STOP/OPEN” button. The running time then
flashes in the display [B] until the operator opens the cover or unlocks it by pressing the“STOP/OPEN” button.

Speed / RCF•

Changes to settings can also be made during a run.

Enter settings by using the arrow buttons [A]. The settings are then shown in the display [A]. Pressing
the arrow buttons once changes the set speed by 50 rpm. When a button is kept depressed, eachchange is increased to a 100-rpm step, while continuous pressure raises this to a 500-rpm step.
The current speed can be shown in g-values by pressing the “RCF” button. The speed can also be
set in g-values by keeping the “RCF” button depressed when pressing the arrow buttons. The “RCF”
steps then correspond to the steps selected for changes to the set speed converted to g-values.

Acceleration Rate••

Changes to settings can be made only when the centrifuge is at a standstill
10 stages are available from 0 (slow) to 9 (fast).

Setting:• Press the “PROGRAM” button.
• Press “Arrow buttons [A]” until the display [A] shows “ACCEL”.
• The display [B] shows the current acceleration rate.
• Make any changes with the “Arrow buttons [B]“.
• Press the “PROGRAM” button to confirm the value set.

V. 3.1 / 07.2006

page E-11

Operating Instruction

Centrifuge LC 24

Braking Rate••

Changes to settings can be made only when the centrifuge is at a standstill
10 stages are available from 0 (slow) to 9 (fast).

Setting:• Press the “PROGRAM” button.
• Press “Arrow buttons [A]” until the display [A] shows “BRAKE”.
• The “braking rate” display shows the current braking rate.
• Make any changes with the “Arrow buttons [B]“.
• Press the “PROGRAM” button to confirm the value set.

Countdown Delay••

Changes to settings can be made only when the centrifuge is at a standstill
When the countdown delay is switched on, the timer starts to count down only when
the rotor has reached the final speed set by the user. This function ensures that the
samples are centrifuged at the selected speed over the entire running time.

Setting:• Press the “PROGRAM” button.
• Press “Arrow buttons [A]” until the display [A] shows “CNTDLY”.
• The display [B] shows the current setting.
• Make any changes with the “Arrow buttons [B]“.
• Press the “PROGRAM” button to confirm the value set.

Sensitivity of the Imbalance Trip•••

Changes to settings can be made only when the centrifuge is at a standstill
This centrifuge is fitted with an imbalance sensor. This ensures that the centrifuge
shuts off in the event of incorrect loading or a fault. This function is for the user’s
safety and extends the life of the centrifuge. The LOW setting allows a greater
imbalance before the centrifuge shuts off.
To achieve maximum life for the centrifuge, set the imbalance trip to “HI” andcentrifuge only balanced loads.

Setting:• Press the “PROGRAM” button.
• Press “Arrow buttons [A]” until the display [A] shows “BALANC”.
• The display [B] shows the current setting (LOW or HI).
• Make any changes with the “Arrow buttons [B]“.
• Press the “PROGRAM” button to confirm the value set.

page E-12

V. 3.1 / 07.2006

Operating Instruction

Centrifuge LC 24

Audible Signals••

Changes to settings can be made only when the centrifuge is at a standstill
A selection can be made from six different audible signals

Setting:• Press the “PROGRAM” button.
• Press “Arrow buttons [A]” until the display [A] shows “BEEPER”.
• The display [B] shows the current setting• Make any changes with the “Arrow buttons [B]“.
Setting

•

1
2
3
4
5

Beep tone whenpressing a button?
NO
YES
NO
YES
NO

6

YES

Signal at the end of the running time
NONE
NONE
YES, 5x BEEPSYES, 5x BEEPSYES, CONTINUOUS BEEP; user must press the“STOP/OPEN” button to stop it
YES, CONTINUOUS BEEP; user must press the“STOP/OPEN” button to stop it

Press the “PROGRAM” button to confirm the value set.

Saving and Retrieving Settings
This centrifuge can save up to 10 user-defined pre-settings. These programs contain all parameters
necessary for determining the conditions for a specific operation (running time, speed, acceleration
rate, braking rate etc.). The user can use the program memory locations to configure the centrifuge
as quickly as possible to a specific type, and to ensure that centrifuging is carried out under identicalconditions by selecting the same program each time.
Saving a setting:• Set all parameters (speed, time, acceleration rate etc.), which are to be saved.
• Press the “PROGRAM” button until the “memory location” display starts to flash.
• Select the desired memory location with the “Memory” button• Press the “PROGRAM” button to save this configuration in the memory location displayed.
Retrieving the setting from the memory:• Press the “Memory” button repeatedly until the desired memory location appears in the“memory location” display.
• As you go through the set parameters, the parameters set with this program appear in therelevant displays.
• When you have found the desired memory program (0-9), wait a short time until the memory
display flashes. The configuration, which was saved is now loaded.
• Once loaded, a configuration remains active until a change is made on the control
panel.
• Memory locations can be overwritten, by making a selection during the storage
process.

V. 3.1 / 07.2006

page E-13

Operating Instruction

Centrifuge LC 24

CounterThis centrifuge is designed for monitoring of device use and routine maintenance oversight by meansof a mechanical counter. The counter is located on the front of the device.

Access to the Rotor Chamber during a Power Failure
If a power failure occurs, it may be impossible to unlock thecover in the usual manner. If this happens, you can gain accessto the rotor chamber by removing the label in front of the coverlock and manually deactivating the locking mechanism with apen (see illustration). Pull the mechanism in the direction of thecontrol panel, unlock it and open the cover. Then re-stick thelabel in position.

If the centrifuge is damaged, have it checked by a technician before further use!

Loading the Rotor••

The centrifuge must be evenly loaded for it to function properly.
For spinning glass- or flat bottom tubes, we recommend, to insert rubber
adapters in the holder base. Otherwise the sample tubes could be damaged.

To ensure even loading, comply with the following rules when loading the horizontal rotor for
six tube holders:
1. Opposite-located tube holders must be identical.
2. Opposite-located tube holders must either be empty or loadedwith the same number of samples of the same weight.
3. When centrifuging an uneven number of samples, insert atube filled with water as a counterbalance.
4. (Only for 4x tube holders): When loading tubes in opposite
locations, a counterbalance is achieved by arranging the tubes
diagonally (not opposite one another).
Example: Loading six tubes in two carriers. In each carrier,two tubes are placed on the base and one tube at top left. See
illustration on the right.

If the tubes are longer than 100 mm (15 ml), do not insert more than (2) tubes per
carrier. Load all these tubes on the same side. For example, to centrifuge twelve (12)15 ml tubes, place two tubes on the right-hand side of each carrier.

page E-14

V. 3.1 / 07.2006

Operating Instruction

Centrifuge LC 24

Centrifuging•

If a problem occurs during a run, which makes it necessary to shut off thecentrifuge, press the “OPEN/STOP” button immediately.

Procedure:
•
•
•
•
•

•
•
•
•
•
•
•
•

Switch on the centrifuge with the main switch.
The main switch lights up.
Press the “OPEN/STOP” button to unlock the cover and open the cover.
Insert sample tubes in the tube holder. Comply with the rules for even loading!
Close the cover and turn the cover knob fully clockwise. The “LOCK” display lights up to
indicate correct locking.
If the cover knob does not fully engage, the “LOCK” display will not light up and thecentrifuge will not operate.
Select the settings desired or load a stored setting.
Press the “START” button.
The rotor starts to rotate. The “RUNNING” display lights up.
At the end of the running time, the “RUNNING” display goes out and the rotor is braked to acomplete stop.
The “UNLOCKED” display lights up, the locking mechanism is deactivated, thereby allowing
access to the rotor chamber. After the automatic unlocking time has elapsed and with therotor at a standstill, the cover can be unlocked with the “OPEN/STOP” button.
Turn the cover knob anticlockwise and open the cover.
Remove the samples.
The centrifuge can be started up again immediately.

••

V. 3.1 / 07.2006

ERROR / BALANC
In the event of an imbalance, the centrifuge will end its current operation andbrake to a complete stop. After the rotor has stopped, open the cover to cancel
the error signal, adjust an even load and start a new run. Alternatively, theerror signal can be cancelled by pressing the "OPEN/STOP” button.
ERROR / SPEED
The centrifuge cannot run at maximum speed if there is a rotor problem,
inadequate power supply or other electrical problems. Press the“OPEN/STOP” button to cancel the error signal and then check the rotor andthe power supply.

page E-15

Operating Instruction

Centrifuge LC 24

8. Maintenance and servicing
Service address
Should you have questions or problems concerning the device, please contact the customer service
department of DESAGA GmbH / SARSTEDT-GRUPPE in Wiesloch or your local SARSTEDT sales
office. Remember to specify the Serial no. of the device.
DESAGA GmbH / SARSTEDT-GRUPPE
Serviceabteilung
In den Ziegelwiesen 1-7
D-69168 Wiesloch
Phone: +49 (0) 62 22 / 92 88 65
Fax:+49 (0) 62 22 / 92 88 60•••

Fill in the Contamination Questionnaire (Page 18) and ship it together with thedevice, a copy of the delivery note, and a short description of the problem.
Always ship the device in the original packaging.
The Contamination Questionnaire serves to protect our service personnel.
Therefore, please fill it in carefully and completely.

Cleaning
Potential danger of contamination

The regulations for cleaning the device must be observed. Inattentive cleaningor non-observance of the regulations can cause malfunctions!
•
•
•
•
•
•
•

Disconnect the device from the mains supply before cleaning.
Disinfect the device before cleaning.
No liquid may be permitted to enter the device during cleaning.
Clean the housing surfaces with a slightly damp cloth. A soap solution may be
used. Carefully dry the device after cleaning.
Do not use scouring agents or aggressive cleaning liquids or solutions (except
alcohol) under any circumstances.
Isopropanol may be used in case of severe soiling.
Fully/partially halogenated hydrocarbons, ketones, esters, and other chemical
not specified by the manufacturer may damage the rotor and tube holders andmay not be used.

Example for a disinfectant:
The following aqueous solution can be used as a disinfectant:
25 g ethanol 96 %, 35 g 1-propanol, 0,1 g glyoxal ad 100 g aqua dest.
Apply the solution, and allow it to react for a few minutes, depending on thecontamination. If necessary, repeat the application of disinfectant.
Tubeholder:
Small glass particles from a tube breakage remaining in the tube holder can stick to
the next sample tube inserted in this holder. When handling this tube, the broken glasscan penetrate protective gloves and cause injury from cuts to the user’s fingers or
hand. Broken glass remaining can result in stress points with subsequent tubes
causing further breakage. If a tube breaks, remove it carefully from the tube holder.
Dispose of the sample and tube fragments correctly, and clean both the inside andoutside of the tube holder. If necessary, use a new rubber adapter and reinsert the
tube holder in the rotor.
page E-16

V. 3.1 / 07.2006

Operating Instruction

Centrifuge LC 24

Regular Maintenance
It is recommended, depending on the amount of use and the external conditions, that the rated speed
and the electrical safety be checked every 1-2 years to ensure continuing safe operation.
It is advisable to exchange the tube holders after 24 months of use.

Index
cleaning..............................................................16
disinfectant .........................................................16
evenly loaded .....................................................14
Machine data........................................................8

V. 3.1 / 07.2006

page E-17

rubber adapter ................................................... 14
safety information................................................. 4
safety switch ........................................................ 7
unusual noises..................................................... 9Operating Instruction

Centrifuge LC 24

Contamination questionnaire
for repair orders
Dear Customer,
Please complete the contamination questionnaire before you return the instrument to the service
department of DESAGA GmbH / SARSTEDT-GRUPPE or to your local SARSTEDT sales
representative for maintenance or repair.

Firm:

Department:

Town:

Street:Name:

Phone No.:

Name of
Instrument/Artikel:

SN:

The instrument is free from dangerous substances
The instrument came into contact with the following dangerous substancesGroups of substances

Substance nametoxic
acridly
explosive
radio-active
contagiously danger substances
biol. hazardous subst.
Other hazardous subst.
The instrument has been decontaminated according to the legal regulations.
(Please cross off if applicable!)
Description of the decontamination:

Date:

Signature:
page E-18

V. 3.1 / 07.2006