CLAMP POWER METER

KEW 2060BT / 2062 / 2062BT

Jaw shape with emphasis on thesafety and the usability
KEW 2062/2062BT

KEW 2060BT

1000A rms
1000V rms
Harmonics up to 30thCurrent up toVoltage up to● Various measurement functions: current, voltage, power, harmonics and phase detection● LCD can display simultaneously the values of voltage and frequency or power and power factor● Each degree of harmonics can be shown on the LCD of the tester● Bluetooth communication function（2060BT and 2062BT only）

Various measurement functions○Measurements of current, voltage,power, harmonics and phasedetection are possible.
○True RMS: Indicates a correct value〜

A AC Current, Frequency
V AC Voltage, Frequency
W Power〜

DATA
HOLDHarmonics

at the distorted waveform.
○Functions of PEAK, MAX, MIN, AVG

are available.
Easy to check the temporal change
in the power source.

DATA HOLD
Phase detection

SETUP○LCD can display simultaneously thevalues of voltage and frequency orpower and power factor.

SET UP

Input terminals canbe used up to 3.
Set the test leads
according to thewiring configuration.

Power measurement
Power measurement on
any wiring system is possible.
* E.g.: 3P4W

S(L2)

T(L3)

Red

LoadSource

R(L1)

N

Black
KEW 2060BT,KEW 2062 and KEW 2062BT canperform Single-phase 2-Wire / Single-phase 3-Wireand balance and unbalance measurements ofThree-phase 3-Wire / Three-phase 4-Wire.
The double display can simultaneously show many
parameters like W & PF, W & deg, W & VA, W &Var, V & A, etc.

Various parameters suchas active/reactive/apparent
power, power factor, phasedifferences(1P2W only)
which are required for thepower measurement canbe measured.

Total 3-phase power canbe measured in simplesteps.

By setting the CT and VT
ratio and measuring thesecondary side of thetransformer, the primaryside value can beobtained.

Harmonics measurement

NLoadL

SourceHarmonics of the voltage and current, which cause
various failures at the field, can be measured.
It is possible to measure and show each voltage andcurrent harmonics up to 30th.
Total harmonics distortion factor is in two types ofTHD-R and THD-F.
Each degree of harmonics can be shown on the LCD ofthe tester.
Both the current value (or voltage value) and the content
ratio can be displayed on the same screen.

Jaw shape with emphasison the safety and the usability
Line up of two types of jaw to meet the test environment○KEW 2060BT has a newly designed special jaw shape forusing at a large busbar. Extremely large jaw withtear drop shape can clamp a large busbar with safe.
(Conductor size 75mm, Busbar 80 x 30mm MAX)

○KEW 2062 and KEW 2062BT have a teardrop shape jaw, and the size is convenient
to use at a small-sized office and factory.
(Conductor size 55mm)

Photo:2060BT

Tear drop shape jaw can keep
the operator free from the dangerof touching the busbar.

Phase detection
Positive phase and negative phase can be conﬁrmed by thebuzzer sound and the display without changing theconnection of the leads.
R（L1）

T（L3）
NBlackYellowRed

LoadSourceS（L2）

KEW Power*
Please search
KEW Power＊
Communication charge may
be incurred separately todownload application.
*Bluetooth® is a trademark or registered trademark ofBluetooth® SIG. Inc. Android™ is a trademark orregistered trademark of Google Inc. iOS is a trademark or registered trademark of Cisco Technology,
Inc. in the United States and other countries.

Download on theBluetooth （ 2060BT and 2062BT only ）○Dedicated Application "KEW Power＊" supports both Android™ and iOS.
○LCD display can be checked at the smart devices.
○Measured values and graphs can be stored with a press of a button.
○Voltage and current can be shown in the waveform display. Existence of theharmonics can be found easy.
○Threshold of each measured value can be set at the application, and PASS andFAIL judgement is possible.

iOS App

Application for iOS
Free iOS software KEW Power＊is available on App Store

ANDROID APP ON

Android ™ App

Application for Android™
Free Android software KEW Power＊is available on Google Play Store● Accessories●KEW 2060BT / 2062 / 2062BT Specifications
2060BT
2062/2062BT
Wiring connections
1P2W, 1P3W, 3P3W, 3P4W
Measurements and parameters Voltage, Current, Frequency, Active power, Reactive power, Apparent power,Power factor (cosθ), Phase angle,Harmonics(THD-R/THD-F), Phase rotation
ACV
1000V
Range
Accuracy±0.7％rdg±3dgt(40.0 to 70.0Hz) ±3.0％rdg±5dgt(70.1 to 1kHz)
1.7 or lessCrest factor
ACA
40.00/400.0/1000A (3 range auto)
Range±1.0％rdg±3dgt(40.0 to 70.0Hz) ±2.0％rdg±5dgt(70.1 to 1kHz)
Accuracy
Crest factor
3 or less on 40.00A/400.0A range, 3 or less 1500A peak on 1000A range
Frequency
Display Range
40.0 to 999.9Hz±0.3％rdg±3dgt
Accuracy
Active power
40.00/400.0/1000kW
Range±1.7%rdg±5dgt (PF1, sine wave, 45 to 65Hz)
Accuracy
Apparent power
40.00/400.0/1000kVA
Range±1dgt against each calculated value, Sum: add errors of each channel, 3P3W: ±2dgt, 3P4W: ±3dgt)
Accuracy
Reactive power
40.00/400.0/1000kVar
Range±1dgt against each calculated value, Sum: add errors of each channel, 3P3W: ±2dgt, 3P4W: ±3dgt)
Accuracy
Power factor-1.000 to 0.000 to +1.000
Display Range±1dgt against each calculated value, Sum: add errors of each channel, 3P3W: ±2dgt, 3P4W: ±3dgt)
Accuracy
Phase angle(1P2W only)-180.0 to 0.0 to +179.9Display Range±3.0°
Accuracy
Harmonics RMS(Content rate)
1st to 30th order
Analysis order
Accuracy±5.0％rdg±10dgt (1 to 10th) ±10％rdg±10dgt (11 to 20th) ±20％rdg±10dgt (21 to 30th)
Total harmonics THD-R/THD-F
0.0% to 100.0%
Display Range
Accuracy±1dgt against the calculated results of each measured value
ACV 80 to 1100V（45 to 65Hz）
Phase rotation
MAX/MIN/AVG/PEAK, Data hold, Bluetooth® (2060BT and 2062BT only), Backlight, Auto power off
Other functions
GeneralCommunication interface Bluetooth®5.0 (2060BT and 2062BT only)
LR6(AAA)(1.5V)×2
Power source
Continuous measuring time Approx. 58 hours
Conductor size
Φ75mm max. (busbar of 80mm×30mm)
Φ55mm max.
Dimension / Weight283(L) × 143(W) × 49(D) mm / Approx. 590g (including batteries) 247(L)×105(W)×49(D)mm / Approx.490g (including batteries)
IEC 61010-1, IEC 61010-2-032, IEC 61326-1, 2-2（EMC）, IEC 60529（IP40）
Applicable standards
CAT Ⅳ 600V / CAT Ⅲ 1000V Pollution degree 2
CAT Ⅳ 300V / CAT Ⅲ 600V / CAT II 1000V Pollution degree 2Accessories
7290 (Voltage test lead set) 9198 (Carrying case) LR6(AAA)×2, Instruction Manual

7290 Voltage test lead set

9198 Carrying case

● Measurement
categories（CAT）
The figure shows an example of measurement
category.
Measuring instruments designed for CAT IV
environments can be used at the environmentsof CAT III, CAT II and CAT O.

Interior wiring
Incoming wire

Socket●Selection Guide of Power Meters
Clamp Power Meter
2060BT

2062

2062BTPower Meter

Power Quality Analyzer

6305

6315

Conductor size
MemoryNumber of Input Channel
Communication interface

An example ofmeasurement
category

CAT Ⅳ300V

Measurement
category

Nominal phase toearth voltage

CautionAppearance

Voltage［V］
Current［A］
Power［W］
Frequency［Hz］
Energy［Wh］
Harmonics
SwellDip
PowerInterruption
QualityTransients
Inrush Current

O: Measuring circuits
without a
MEASUREMENTCATEGORY.

✓✓
✓
✓

✓
✓
✓
✓

✓
✓
✓
✓

✓

✓

✓

Φ75mm
4ch（V3,A1）
Bluetooth®

Φ55mm
4ch（V3,A1）

Φ55mm
4ch（V3,A1）
Bluetooth®

✓
✓
✓
✓
✓

✓
✓
✓
✓
✓
✓
✓
✓
✓
✓
✓

Differs depending on theDiffers depending on theoptional clamp sensors used. optional clamp sensors used.

SD card6ch（V3,A3）
USB,Bluetooth®

●This leaflet was produced in Novembar 2023. The contents of this leaflet are subject to change without notice.

SD card7ch（V3,A4）
USB,Bluetooth®

The voltage value which follows after the category means the limit of the phase to earth
voltage. It is not the limit of the phase to phasevoltage. Therefore, in the case of a measuring
instrument which complies with CAT IV 300V,
it can be used at the electrical circuit up to 520V
phase to phase at the 3-phase line connected bystar connection.

Some countries regulate the com-pliance with their Radio Law of theproducts equipped with Bluetooth .

Please conﬁrm it with your distributor
before purchasing our products equipped with Bluetooth .

LC̲2060BT̲2062̲2062BT̲2̲0̲Nov23̲SS̲en