PSA Oxygen Concentrator System

Regalia

Regalia™ oxygen concentrators offer thesimplicity of the Advanced Technology
Fractionator® (ATF) oxygen module in asystem that is quiet, lightweight, and easilytransportable, making it a perfect fit inapplications in which the concentrator may
be next to the consumer or user. Regalia
concentrators are ideal for applications that
do not require a prescription, but can benefit
from the sound reduction of plastic housing, theconvenience of a handle and casters, the uniqueelegance of the EZ interface, and the reliabilityof the ATF technology.
With the EZ interface, simply set the desired
flow and the oxygen concentrator will make thenecessary adjustments to meet that flow. If theunit is turned off, it returns to the most recent
setting when the unit is next turned on. If adownstream flow constriction like a flow meter or
torch valve constricts the flow, it is no problem
for the Regalia. When the constriction isremoved, the Regalia returns to the target flow.

Features■■ Maintenance-free ATF oxygen module■■ Valve purge noise eliminated for quiet

operation■■ Easy-to-read backlit flow display■■ Simple keypad controls: On/off and O2
increase/decrease arrows■■ Oxygen concentration monitor(with 3-light display)■■ Locking casters

Typical Applications
Manufacturing■■ Glass Work/BlowingMedical■■ Ozone Therapy■■ Health Spas and Clubs
Additional■■ Veterinary■■ Hyperbaric■■ CosmeticSpecifications

12

Performance Data1
Product Flow2
Oxygen Delivery Pressure

21 SCFH (0.55 Nm3/hr or 10 LPM
7 psig (48 kPa or 0.48 barg)

Product Concentration

90% +/- 3%

Product Dew Point-100˚F (-73˚C)

Dimensions (W x D x H) (nominal)

14.7 x 19.5 x 26.5 in (37 x 50 x 66 cm)

Weight57 lb (26 kg)

Physical Connections
Product Gas Outlet

DISS fitting

Storage Temperature-4ºF (-20ºC) to 140ºF (60ºC)

Ambient Operating Conditions

Locate the oxygen concentrator system in a well-ventilated area that is protected from weather elements
and remains between 50°F (10°C) and 104°F (40°C) inside room

Control Power Requirements (Single Phase)

115 V ~ ±10%, 60 Hz, 5.0 A or 220 – 240 V ~ ±10%, 50 Hz, 2.4 A

Typical Power Consumption

600 W

AlarmsPower failure; O2 concentration (optional)

Outlet pressure may decline to 5 psig (34.5 kPa) nominal at 10,000 ft altitude. The system performance may decline above 10,000 ft (3048m).
SCF (Standard cubic foot) gas measured at 1 atmosphere and 70°F / Nm3 (Normal cubic meter) gas measured at 1 atmosphere and 0°C / LPM (Liters per minute) gas measured at 1 atmosphere and 21°C

The ATF Advantage

ATF oxygen modules incorporate proven pressure swing adsorption (PSA) principles into a unique patented
design, which allows these systems to be compact, efficient, rugged, and lower in cost. The ATF module offersunparalleled design flexibility and enables applications where on-site/on-board oxygen generation was previouslyimpractical.
A patented single rotary distribution valve built into the ATF module is continuously rotated at low speed by a small
motor. The valve is maintenance free, self-cleaning, insensitive to contamination, and invulnerable to wear. It
sequentially directs the flow of compressed air to a group of four sieve beds (adsorption), while at the same time
another four beds are purged into the atmosphere through the valve (desorption). The remaining four of the twelve
beds are interconnected through the valve to equalize pressure as the sieve beds sequentially transition between
adsorption and desorption. The small amplitude pressure swings generated by the ATF’s twelve sieve beds
eliminate loud noise pulses, eliminate the need for a pressure regulator, and reduce compressor wear.

Air inO2 out

26 [66]15 [38]ModelPart Number

Regalia

Description

19 [48]Ordering Information

5161-SEQ

With oxygen concentration monitor (LED indicator) 115 V ~ ±10%, 60 Hz

5162-SEQ

115 V ~ ±10%, 60 Hz

5163-SEQWith oxygen concentration monitor (LED indicator) 220 – 240 V ~ ±10%, 50 Hz

5164-SEQ

220 – 240 V ~ ±10%, 50 Hz

Shipping Information

Regalia

Class92.5

Commodity Classification Number

8421.39.8040

Dimensions (W x D x H)

24 x 20 x 30 in (61 x 51 x 76 cm)

Gross Weight

67 lb (30 kg)

Warranty: 1 Year Parts and Factory Labor** An unprotected or inadequately ventilated environment, or improper control power may cause damage to the oxygen generator 		
not covered under warranty.
All performance ratings based on an ambient temperature up to 100°F (38°C), up to 1,000 feet elevation, and 80% relative humidity.

MADE IN USA© 2019 AirSep Corporation. All Rights Reserved.
AirSep reserves the right to discontinue its products, or change the prices, materials, equipment, quality, descriptions,specifications and/or processes to its products at any time without prior notice and with no further obligation orconsequence. Any rights expressly stated herein are reserved by us, as applicable.

260 Creekside Drive
Buffalo, NY 14228-2075 U.S.A.
Tel: (716) 691-0202 ■ Fax: (716) 691-1255
www.airsepcpd.com ■ info@airsep.com
ML-IND0026 E