HIGH STIFFNESS / HIGH RESPONSE TYPE LOAD CELL

Flexible cable"Virtually unbreakable!" - Loadcell with an innovative concept!
A strain gauge sensor with phenomenal rigidity and responsiveness.
RobustHigh responsivenessAmazing rigidity is achieved with a strain gauge!
As it rarely gets deformed when loaded, Super Cell can be usedfor variable applications such as measurement of grinding
power of machine tools.

High resonant frequency is possible with strain gauge typesensor!
It can be used to monitor a high-speed phenomenon.

Safe overload of 500%
Overload up to 500% of the rated capacity does not affect theaccuracy.
Stable measurement can be performed for a long term even insever loading conditions (e.g. press fitting machines).

Example of use
PLC
Press machine

STARTetc.

OK/NOK
judgment etc.

Displacement sensor

High stiffness &high response loadcell

SD cardLoad cell

Indicator

USBPC

Specifications
Model● Wiring diagram

SCB-□N

Rated capacity

100 N 200 N 500 N

Rated output

0.4 mV/V±0.2%*1

1 kN

2 kN＋ EXC (Red)□ : Rated capacity

5 kN

10 kN 20 kN 50 kN 100 kN 200 kN
0.4 mV/V±0.1%

Safe overload±1% R.O.*1

0.2 or less R.O.

Hysteresis

0.2 or less R.O.

Repeatability

0.1 or less R.O.

Compensated temperature range－10 to +60℃

－ SIG (White)
SHIELD (Blue)

Safe temperature range－20 to +70℃

Temperature effect on zero

0.015% R.O./℃ or below

Temperature effect on span

0.01% R.O./℃ or below

Input resistanceApprox. 700 Ω

Approx. 1000 Ω

Output resistanceApprox. 700 Ω

Approx. 1000 Ω

Recommended excitation voltage

5V

10 V

Maximum excitation voltage

7.5 V

15 V

Insulation resistance (DC 50 V)

1000 MΩ or more
φ2.5 6-conductor flexible cable φ5 6-conductor flexible cable
3 m, Cable end: 7 wires
3 m, Cable end: 7 wires

Cable(Orange)＋ SIG (Green)±0.3% R.O.*1

Non-linearity(Yellow)－S－ EXC (Black)500% R.C.

Zero balance＋S

φ7 6-conductor flexible cable 3 m
Cable end: 7 wires

Loadcell material

Stainless steel

RoHS directives

2011/65/EU (EU)2015/863*1 Specification changed.

External dimension■ SCB-100N/200N/500N■ SCB-1KN/2KN/5KN

Load direction
(+)

SR50

φ12SR75

25222118

8.5(3)

φ10(3)

(4.5)

SR30

910(1)

φ4■ SCB-10KN/20KN

Load direction(+)

Load direction
(+)

8φ20
φ60

φ40

P.

C
P.

P.C.D.17

C.

.3.D

2-M2 Depth 3.590°

3D.

48
22.5°

4 5°

φ2.5 6-conductor
flexible cable 3 m
Cable end: 7 wires *67. 5°

2-M2.5 Depth 4

φ5 6-conductor
flexible cable 3 m
Cable end: 7 wires *φ7 6-conductor
flexible cable 3 m
Cable end: 7 wires *4-M4 Depth 6■ SCB-100KN

SR150

φ5035φ100CP.
.8.D°

P.

0C.

D.

80

2-M6,effective
depth 12
Not used to fix loadcell main body

ModelDeflection at rated capacity Natural frequency Weight
20 g
0.015 mm
40 kHzSCB-100N
20 g
0.015 mm
40 kHzSCB-200N
20 g
0.015 mm
40 kHzSCB-500N
130 g
0.015 mm
26 kHzSCB-1KN
130 g
0.020 mm
26 kHzSCB-2KN
130 g
0.025 mm
26 kHzSCB-5KN
400 g
0.025 mm
26 kHzSCB-10KN
400 g
0.035 mm
26 kHzSCB-20KN
2.0 kg0.065 mm
15 kHzSCB-50KN
2.3 kg0.100 mm
15 kHzSCB-100KN
8.2 kg0.100 mm
8 kHz
SCB-200KN

φ154°

φ7 6-conductor
flexible cable 3 m
Cable end: 7 wires *9 0°

P.C.D.110

22.5°

φ7 6-conductor
flexible cable 3 m
Cable end: 7 wires *

90

22.5°

2 2.5°

4-M5 Depth 8SR200

20

φ1009064

1750(5)

φ24

Load direction
(+)

45

1340

SR15035(5)φ24Load direction
(+)

（6）

Load direction(+)■ SCB-200KN70■ SCB-50KN

φ7 6-conductor
flexible cable 3 m
Cable end: 7 wires *4-M5 Depth 10
4-M8 depth10,pilot hole depth14
Equal circumference

Unit: mm

* Connector can be attached to the cable according to your requests.
Refer to 82 pages for details.