FC400-CCL
FC400-DAC
FC400-EIP
FC400-ECT
FC400-232
DIN-RAIL MOUNT WEIGHING INDICATOR
NDICATOR
It is suitable for many weighing applications such ashopper scale, packing scale, weight level meter, etc….
FC400-CCL
FC400-DAC
FC400-EIP
FC400-ECT
FC400-232

A weighing system can be configured easily using CC-Link network.
Either voltage or current output, corresponding to the indicated value, is available.
A weighing system can be configured easily using EtherNet/IP network.
A weighing system can be configured easily using EtherCAT network.
Measured data and results can be collected using RS-232C network.

High sampling rate & resolution

Various interfaces

High-Speed A/D conversion and powerful digital processing
capability of 1200 times/sec.
High display resolution of 1/100000 (max).

FC400-CCL Equipped with CC-Link, SI/F, USB interface
FC400-DAC Equipped with D/A converter, RS-485, USB interface
FC400-EIP Equipped with EtherNet/IP and USB interface
FC400-ECT Equipped with EtherCAT and USB interface
FC400-232

Equipped with RS-232C, SI/F, USB interfaceex) CC-Link
EtherNet/IP
EtherCAT
RS-485

Measurement can be performed quickly andprecisely due to high speed A/D conversion.

Application software for USB interface
With communication through USB
interface, logging, graphdisplay, setting parameters, andcalibration can be done.

CC-Link
EtherNet/IP
EtherCAT
RS-485

Load cellinput

FC400

High performance filter & auto filter adjustment
PLC

With combination of low pass filter (0.1 to 300 Hz) whichcorresponds to various vibration and moving average filter (OFF,2 to 512) that is effective for periodic vibration, automatic
searching for optimal value with accuracy and stability
can be achieved. (You can manually adjust the settings too)The
readings
fluctuate….

By simply
pressing
the autoadjustment
key….

ex) CC-Link
EtherNet/IP
EtherCAT
RS-485
RS-232C
D/A converter
SI/F

Sensor
input
SI/F

Display etc.

D/A
FRQYHUWHU Recorder
Stabilized!

RS-232C

Compact size
FC400

It can be fit into a limited space!
Light weight & compact body for 35 mm DIN rail mount.
ex) USB

CC-Link
EtherNet/IP
EtherCAT
RS-485

PLC

FC400
USB

PC

Basic weighing process control function

Equivalent input calibration

Equipped with weighing sequence function to control
feeding/discharge gate.

Theoretical calibration can be performed easily by registering thecapacity and rated output of load cells.

Input conversion value display

Multipoint calibration (linearization)

The output signal level of load cell can be displayed in mV/V for
monitor purpose.
Malfunction indicator or faulty sensor can be differentiated easily.

Three additional points can be defined in the middle between
zero and span for better linearity. Eventhough the scale has poor
linearity, it can be corrected to be a highly accurate scale.

6-digit display
24 bit A/D converter enables 6-digit display

Specifications
GeneralPower supply voltage
specification Power consumption
Operating conditions

Dimensions
WeightDC 24 V (±15%)
3 W typ. (FC400-DAC, FC400-232)
4 W typ. (FC400-CCL, FC400-EIP, FC400-ECT)
Operating temperature range: -10 to +50Υ
Storage temperature range: -20 to +85Υ
Humidity: 85% RH or less (non-condensing)34(W) × 88(H) × 91(D) mm (Projections excluded)
Approx. 210 g (FC400-CCL, FC400-DAC)
Approx. 230 g (FC400-EIP, FC400-ECT)
Approx. 220 g (FC400-232)

Quick manual…2, Jumper wire…2, Mini driver…1, CC-Link connector…1 (FC400-CCL)
Various I/O connector (FC400-CCL, FC400-EIP, FC400-ECT)…2,(FC400-DAC, FC400-232)…3
Accessories CA81-USB: miniUSB-computer USB cable 1.8 m
CN74:
CC-Link connector (Same accessory as the attached one)
CN75:
CC-Link connector (Y type branch connector)
CN76:
CC-Link connector (Terminator connector)
CN85:
13 p connector for power source/sensor/RS-485(Same accessory as the attached one)CN86:
3 p connector D/A converter (Common with RS-232C connector)(Same accessory as the attached one)CN87:
10 p connector for external I/O (Same accessory as the attached one)

Attachment

CE marking
certification

EMC directive EN61326-1* FC400-ECT is Free Run mode.
* Please note that there are possibilities of individual differences in a color tone on display devices suchas LEDs, fluorescent display tubes and LCDs due to manufacturing process or production lots.

Structure of product code
Model
FC400-CCL
FC400-DAC
FC400-EIP
FC400-ECT
FC400-232

Interface
CC-Link, SI/F, USB
D/A converter (Voltage and current output), RS-485, USB
EtherNet/IP, USB
EtherCAT, USB
RS-232C, SI/F, USB

External dimension

DAC

DIN-rail mountadapter‫ ڦ‬FC400-EIP

88101

88101
91

34

34

‫ ڦ‬FC400-232

DAC-FA‫ ڦ‬FC400-DAC

CCL-FACCL‫ ڦ‬FC400-CCL

91

34

DIN-rail mountadapter

34

DIN-rail mountadapter

91‫ ڦ‬FC400-ECT

88101

EIP-FA

Interface

EIP

Externalsignal

88101

Display

DC 5V±5% Output current: Within 90 mA
DC 2.5V±5% Output current: Within 45 mA(depending on settings)
Ratiometric method(Up to 6 350 Ω load cells can be connected in parallel)-2.5 to +5.1 mV/VSignal input range
Zero adjustment range Automatic adjustment by digital processing -0.5 to +2.0 mV/V
Span adjustment range Automatic adjustment by digital processing 0.01 to 3.0 mV/VLinearization function Up to three point multi-point calibration is possibleusing linearization function
Min. input sensitivity
0.15 μV/countAccuracyNon-linearity: Within 0.01% FS
Zero drift:
0.0002% FS/Υ typ. * When calibrated by 3 mV/V with 5 Vof the excitation voltage
Gain drift:
1 ppm/Υ typ.
Digital low-pass filter 0.1 to 300 HzFilter
Moving average filter OFF, 2 to 512 times
Speed:
1200 times/sec.
A/D converter
Resolution: 24 bit (binary)Character height 8 mm
Display unit
Numerical display by 7-segment green LED
Up to 6 digits.
Display value
Selectable from 1, 3, 6, 13, 25 times/sec.
Display frequency
RUN, SD, RD, ERR (FC400-CCL)
Status display
MS, NS (FC400-EIP)
RUN, ERR (FC400-ECT)
Transistor’s open collector output
Output signal(5)
Vceo = 30 V, Ic = 50 mA
Input signal
Selectable/configurable(3)
Input is ON when shorted to COM terminal by contact (relayswitch, etc.) or non contact (transistor, open collector, etc.).
* DC24 V external power supply is required.
CC-Link interface (FC400-CCL)
D/A converter (Voltage and current output) (FC400-DAC)
EtherNet/IP interface (FC400-EIP)
EtherCAT interface (FC400-ECT)
RS-232C interface (Selectable from Modbus-RTU or UNI-Format) (FC400-232)
RS-485 interface (Selectable from Modbus-RTU or UNI-Format) (FC400-DAC)
SI/F 2-wire type serial interface (FC400-CCL, FC400-232)
USB interface
Excitation voltage

88
101

Analog91

DIN-rail mountadapter

3491

DIN-rail mountadapter

Unit: mm