LASER BEAM MEASUREMENT& BEAM PROFILINGPRODUCT GUIDE 2024
POWER

ENERGYPROFILINGTERAHERTZNEW RELEASES
Featured products

t

c
du

ew

PRONTO-250-FLEX
The popular PRONTO-250 is now available with FLEXible calibration options, so you only
pay for the features you need.

o
pr

N

THE BASIC MODEL INCLUDES A BROADBAND CALIBRATION:• Default calibration “Y”: for visible to NIR wavelengths (248 nm to 2.5 µm)
OPTIONAL CALIBRATIONS YOU CAN ADD:
• Additional calibration “C”: for CO2 lasers (10.6 µm)• Additional calibration “E”: for energy measurements with ± 5 % accuracy

PORTABLE & EASY TO USE

t

uc

ew

od
pr

QE195 : THE LARGEST ENERGY DETECTORS
Custom-built to your specifications, contact us with your laser measurement needs

N

TWO SIZES AVAILABLE, FOR EXTRA-LARGE BEAMS:• 145 mm Ø

• 195 mm Ø
MEASURE HIGH ENERGY: UP TO 700 J PER PULSE, DEPENDING ON PULSEDURATION

THE LARGEST PYROELECTRIC DETECTORS ON THE MARKET

t

uc

ew

od
pr

IS50 : ENERGY METER FOR HIGH AVERAGE POWER
Custom-built to your specifications, contact us with your laser measurement needs

N

DESIGNED FOR HIGH ENERGY MEASUREMENTS AT HIGH REPETITION RATESCAN HANDLE UP TO 1000 W AVERAGE POWER
OUR PROPRIETARY COATING OFFERS DAMAGE THRESHOLDS THAT ARE
ORDERS OF MAGNITUDE
HIGHER THAN ANY OTHER “WHITE” COATING ON THE MARKET.

PULSE-TO-PULSE ENERGY AT UP TO 1 KW AVERAGE POWER

ct

ew

N

u
od
pr

HP100A-50KW : SMALL BEAM, EXTREMELY HIGH POWER
With its gold reflector cone and back-reflection reducing geometry, the HP100A-50KW-GD
traps > 97 % of the incident light and can handle up to 50 kW of continuous laser power.
The very low back-reflections of the detector ensure a safer working environment withhigh-power lasers. This detector can also handle the high intensities of small beams.
HIGH POWER, LOW BACK-REFLECTIONSTABLE OF CONTENTSABOUT GENTEC-EO

4

POWER DETECTORS

12LOW POWERPH
PRONTO-SI
UM-B
XLP12
GENERAL USE
UP10-H
UP12-H
UP17-H/W
UP19-HPRONTO
UP25-H
UP55-H
UP55-HD
HIGH-PERFORMANCE ABSORBERS
UP19-W
UP50-W
UP16-QED
UP52-QEDIS
HIGH POWER
HP60
HP100
SUPER HP
HIGH-POWER PRONTO
BD
ABSORPTION CURVES

16

16
18
20
22

24

24
26
28
30
32
34
36
38

40

40
42
44
46
48

50
50
52
54
56
58

60

ENERGY DETECTORS64

LOW ENERGYPE
QE-B
MACH 6
GENERAL USE
QE12-MB
QE25-MB
QE50-MB
QE65-MBQE95-MBQE145-MB
QE195-MB
HIGH SPEED
QE12-MT
QE25-MT
QE50-MT
HIGH ENERGYIS50PRONTO-500-IPL
CALORIMETERS
ABSORPTION CURVES

68

68
70
72

76

76
78
80
82
84
86
88

90

90
92
94

96

96
98
100

104

BEAM PROFILING

106

BEAM PROFILERS
BEAMAGE108

PRODUCT GUIDE 2024

108

QUALITY ANALYZERSBEAMAGE-M2
POWER MANAGEMENTNDBAWAVELENGTH MANAGEMENTUV CONVERTERS
IR ADAPTORFILTERSBEAM SIZE MANAGEMENT
CL
ABSORPTION CURVES

112

112

116
116
118

120

120
121
122

123

123

124

TERAHERTZ DETECTORS126

LOW POWER
THZ-B
THZ-I-BNC
GENERAL USE
THZ-D
SENSORSQS-THZ
ABSORPTION CURVES128

DISPLAYS & PC INTERFACESDISPLAY DEVICESMIRO ALTITUDE
MAESTRO
MAESTROTUNER
UNO
PC INTERFACES
U-LINK
S-LINK
P-LINK
M-LINK
ALL-IN-ONE DETECTORSINTEGRAPC-GENTEC-EO
BLU

CUSTOM / OEM PRODUCTS
OEM SENSORSUDDISCRETE PYROSQS-L & QS-H
QS-IF
QS-IL
DISCRETE PYROSCUSTOM PRODUCTSCUSTOM DESIGN EXAMPLES
QUAD
STEP
TRAP

128
132

134
134

136
136

138

140
144

144
150
152
154
156

158

158
160
162
164

166

166
167
168

170
172

172
174
175
176
177
178

180

180
182
186
188

DISTRIBUTORS WORLDWIDE190

RETURNS & WARRANTY

191

T 418.651.8003 | info@gentec-eo.com

.com

3

ABOUT GENTEC-EO
Partners for accuracy

Located in the heart of the Quebec Optical Hub, in beautiful Quebec
City, Canada, Gentec Electro-Optics (Gentec-EO) has a long history in
the laser measurement field. With a track record of over 50 years ofinnovation and providing quality solutions for laser power and energy
measurement applications from the factory to the hospital andlaboratory, Gentec-EO stands ready to serve you now and in the
future.

4

gentec-eo.com

MILESTONESThe first laser energy meter in the world has been initially developed for internal use as
Gentec Inc. were putting the first high repetition rate TEA CO2 lasers on the market
in 1970. Gentec, Inc. introduced the first pyroelectric joulemeters shortly after that.
They were also the first to manufacture both thermopile wattmeters and pyroelectricjoulemeters. In the mid 1990’s, Gentec introduced the WB series with an average
power density damage threshold of 100 kW/cm² that is still unrivalled today. In 2000,
Gentec Electro-Optics, Inc. was formed from Gentec, Inc. so that the focus was entirelyon laser measurement. And in 2010, the acquisition of Spectrum Detector Inc. allowed
Gentec-EO to cover new markets, like THz detectors, ultrafast pyroelectric detectors
and highly sensitive photodetectors, to name a few.

OUR ESSENCE
The decision of adopting “ PARTNERS for ACCURACY ” as our branding slogan is theresult of a long evolution that spanned over more than 50 years. It came to us naturally
since it represents our very essence. We have always aspired to be more than a simple
supplier of state-of-the-art laser measurement technologies. We truly believe thatdeveloping a very close partnership with our customers is essential and beneficial
for every party. By definition, “ partnership ” means “ aiming at the same goal ” and“ working together ”. This is what is driving us. As for “ accuracy ”, it does not solely
refer to the precise measurements we are able to provide, but also to the complete
understanding of our customers’ needs and expectations. Finally, the key to our
success is to focus all our energy into “ rigor ”. No matter what the situation, Gentec-EO
is always proud to offer its customers the most accurate laser measurements as well as
the most personalized help for the development of custom products and solutions.
Let us be your PARTNERS for ACCURACY.

WORLDWIDE PRESENCE
Gentec-EO has an evergrowing presence everywhere around the world. We currently
have partners in over 40 countries, and each year, we keep adding new partners. We
also have a strong presence in most of the European and Asian countries and we now
have offices in USA and in Japan. When you send a unit to us for repair or recalibration,
you are entitled to expect your unit back in as short a time as possible.
With calibration centers on 3 continents, and offices in Canada, USA and Japan,
Gentec-EO has a solid presence and fast turnaround times, just what you need to keep
pace with today’s rapid market.

PRODUCT GUIDE 2024

T 418.651.8003 | info@gentec-eo.com

.com

5

HIGHEST CALIBRATION STANDARDS
Measuring with Gentec-EO accuracy

At Gentec-EO, we understand that the essence of our business sinceover 50 years has been delivering accuracy. There are no half measures :it either measures accurately or it doesn't. This is why one of ourcompany's values is rigor, because our customers expect nothing less.

6

gentec-eo.com

THE GENTEC-EO ADVANTAGE

We use only GOLD Calibration Standards,
guaranteeing our customers the lowest calibration
uncertainty possible
For each detector that we calibrate,

50 Parameters are collected and logged

Our uncertainty values are based on

Proven Statistical Calculation Processes
Our Personnal Wavelength CorrectionTM (PWC) dataoffers you NIST and/or NRC Traceability overin our ISO-certified quality system

the entire range of the detector

The calibration reference is checked
2 to 3 Times during EACH calibration process

TOTAL ACCURACY of your detector

Each of these steps contributes to the

THE TERMS

ACCURACYUNCERTAINTYERRORThe accuracy of a measurement
is defined as the closeness of theagreement between the result of ameasurement and the true value.

Uncertainty is a measure of the“goodness” of a result. The definitionand concept of uncertainty is aquantitative attribute to the final
result of measurement, considering
all systematic and random
components of all known input
quantities.

The error on a measurement
is the difference between themeasurement result and the true
value.

REPEATABILITY

REPRODUCIBILITY

PRECISION

The repeatability is the closenessof the agreement between the
results of successive measurements
under the same conditions ofmeasurements.

The reproducibility is thecloseness of the agreement
between the results of successive
measurements under changed
conditions of measurements. Thisis also defined as “precision under
reproducibility conditions”.

The precision of a measurement
is defined as the closeness ofagreement between independent
test results obtained under stipulated
conditions.

PRODUCT GUIDE 2024

T 418.651.8003 | info@gentec-eo.com

.com

7

HIGHEST CALIBRATION STANDARDS
Measuring with Gentec-EO accuracy

THE CALIBRATION PROCESS

1Comparison
Process

4 Certification

ACCURACY

2 GoldStandard

8

gentec-eo.com

3 Uncertainty
Calculation

THE TECHNIQUE

By definition, calibration is a comparison between measurements, one of a known magnitude or correctness, which istypically called a “gold standard”, and another measurement comparable to the first one. In the calibration process, there are
four critical aspects that need to be controlled precisely:

COMPARISONPROCESSCERTIFICATION

The first step in the calibration process is thecomparison to a known and traceable standard. At
Gentec-EO, we always do this using Gold and not
a Silver calibration standards, unlike some of ourcompetitors. This extra carefulness in the comparison
process comes from decades of experience in the laser
measurement business.

Gentec-EO has been using its own control quality
system for many years and is now certified ISO
9001:2015 and our calibration laboratory is accredited
ISO/IEC 17025:2017. Over and above the quality system
certification process, the most important aspect is how
rigorously the different steps and parameters arecontrolled in order to deliver an accurate calibration
day after day.

GOLD
STANDARD

UNCERTAINTYCALCULATION

Gentec-EO’s gold laser power detector heads are compared to
NIST standard calorimeters at different wavelengths, in accordance
to the different lasers used to calibrate your own detector heads.
The laser beam has a nominal diameter appropriate for thedetector, and is centered on the detector’s absorbing surface. The
laser energy impinging upon the test instrument is measured
concurrently using a NIST standard calorimeter and a calibrated
beam splitter. The beam splitter ratio is measured using NIST
standard calorimeters. Before the measurements are performed,
the test instrument is allowed to reach equilibrium with thelaboratory environment. The calibration factor is found by dividing
the instrument output reading by the calculated average incident
laser power. The calculation is based on the output reading of the
NIST standard calorimeters.

At Gentec-EO we offer the best uncertainty on themarket, which means more than just giving thecustomer the lowest uncertainty value. These
calculations also need to follow recognized statistical
calculation standards, including those given in NIST’s
Technical Note 1297. Another important parameter toverify, and one that is less known, is the confidence
level. At Gentec-EO, we use a very high confidence levelof 95 %. Like every other step in the calibration process,our uncertainty calculations are done rigorously. We
don’t aim to give you just the lowest number, whateverits meaning, we rather aim to give you the true value,with the highest confidence possible.

PRODUCT GUIDE 2024

T 418.651.8003 | info@gentec-eo.com

.com

9

HIGHEST CALIBRATION STANDARDS
Measuring with Gentec-EO accuracy

ELECTRICAL INSTRUMENTS

All of our electrical instruments are calibrated by certified calibration suppliers. They certify that, at the time of calibration, theinstruments used for calibration meet or exceed all published specifications and have been calibrated using standards whose
calibrations are traceable to the NIST and/or other recognized international standards. The electrical and physical properties of
their laboratories meet the highest requirements for ambient temperature, relative humidity and cleanliness. Their equipment ismaintained by procedures that meet the requirements of ISO 9001:2015 and ISO/IEC 17025:2017.

THE FACTS
HOW GENTEC-EO CALIBRATES YOUR DETECTOR

Every detector is individually calibrated to the best possible accuracy traceable to NIST standards. Stable laser sources at various
wavelengths are used in our calibration process.

UNCERTAINTY

One very common misconception is the absolute value of calibration uncertainty. Be aware that this value is made using a complex
statistical method that takes in account ALL the sources of uncertainty that are present in the process. The figure below shows thesesteps and their respective contribution to the value of uncertainty. As you can see, the manufacturer itself is only one of these sources.

-3%

-2.5%

-2%

-1.5%

-1%

-0.5%

+0.5%

+1%

+1.5%

+2%

True Value

NIST Reference (±0.3% to ±0.9%)

Gentec-EO Gold Standard (±1%)(Calibrated by NIST)

Gentec-EO
Calibration Setup±1.5%

Your Gentec-EO Detector (±2.5%)(Calibrated by Gentec-EO using a Gold Standard)

10

gentec-eo.com

+2.5%

+3%

CALIBRATION WAVELENGTHS

Another misconception is that any wavelength can be NIST calibrated. The NIST only supplies references for distinct wavelengths
contained between 157 nm (F2 excimer lasers) and 10.6 μm (CO2 lasers). Every other wavelength within this range or out of this range issubject to an additional error.
For more information about NIST’s calibration wavelengths, please visit their website at: https://www.nist.gov/calibrations

PERSONAL WAVELENGTH CORRECTIONTM CERTIFICATE
To fill the gaps between the NIST references, Gentec-EO offers you the only NIST traceable calibration in nm steps, from 250 nm to 2.5µm. We achieve this using our proprietary setup that is based on a NIST traceable spectrophotometer. This way, instead of supplying
you with typical values, we offer you a NIST traceable calibration. What you get is an overall accuracy that is not more than ± 1 % awayfrom the original calibration accuracy, in the calibrated spectral range.
Each Gentec-EO detector comes with a Personal wavelength correctionTM Certificate. The correction factors are based on
measurements that were made with YOUR detector. They are not based on the general curve of the absorbing material or the general
response of equivalent products. This means you get the best wavelength correction tool available on the market. This data is stored in
the smart interface of your Gentec-EO detector, you just have to select the wavelength in your display device or PC interface to get themost precise laser measurements on the market.

Your
detector model

TM

Personal wavelength correction

Certificate

Spectral Absorption Plot measured for: UP55N-40S-H9-D0 Power Detector

Serial #299999

Personal Wavelength Correction TM

Absorption curveof your detector

Wavelength***(nm)

Correction
Multiplier
Uncertainty

193
213

0.958

N/A

0.950

N/A

0.949

± 2.5 %

0.958

± 2.5 %

308

0.950

± 1.0 %

337

0.948

± 1.0 %

355

0.949

± 1.0 %

488

0.942

± 1.0 %

514

0.946 3

532

0.949

± 1.0 %

578

0.956

± 1.0 %

632

0.965

± 1.0 %

694

0.974

± 1.0 %

248
266

1

± 1.0 %

720

0.979

± 1.0 %

810

0.991

± 1.0 %

1.000

± 1.0 %

1.000

N/A

980
1064 *

20.978± 1.0 %

2100

0.953

± 1.0 %

10600**

1550

0.944

N/A

Wavelength
correction

1 Wavelengths programmedin the EEPROM (nm)(Based on the absorption
curve of your detector)

2 Calibrated wavelength(nm)(Using a gold standard)

3 Correction factors(Multipliers)* Calibration wavelength** Typical value
Adjustment multiplier for wavelength under 248 nm are nottraceable.
For Gentec-EO monitors, select the proper wavelength in menu
For other monitors, multiply by the correction multiplierPower corrected = Power read x correction multiplier
Example: Power (488 nm) = 10mW x 0.942 = 9.42 mW

For more info, see Application Note 202184 - Understanding your Calibration Certificate.

PRODUCT GUIDE 2024

T 418.651.8003 | info@gentec-eo.com

.com

11

POWER DETECTORS
Low power measurement

Available with

PH: PHOTODETECTORS••••

Photodetectors for measurements up to 750 mW
Available from UV to IR
Silicon, UV-silicon and germanium sensors
OD.3/OD1/OD2 attenuators available

FAST RESPONSE POWER DETECTORSPRONTO-SI: ALL-IN-ONE PHOTODETECTOR + METER••••

Compact laser power meter up to 800 mW
10 x 10 mm aperture
Integrated OD1 slide-in attenuator
Color touchscreen display

PORTABLE & EASY TO USE

UM: BROADBAND PYROELECTRIC DETECTORS
Our pyroelectric power detectors have the noise level of a photodetector, but withthe large bandwidth of a pyroelectric sensor. They have everything you need toaccurately measure extremely low powers from the DUV to the FIR.
• 9 mm Ø aperture• Broadband, flat spectral response• Very low noise, down to 5 nW

MEASURE LOW POWER AT ANY WAVELENGTH

Available withXLP12: LOW-POWER THERMOPILES• Low noise level: only ±0.5 μW• IR filter available• Available with volume absorber for short pulses

THERMAL POWER DETECTORS WITH LOW NOISECOMPARISON TABLE - LOW POWER MEASUREMENT
Spectral range
0.35 - 1.08 µm
0.21 - 1.08 µm
0.80 - 1.65 µm
0.32 - 1.1 µm
0.10 - 20 µm
0.19 - 20 µm

PH100-Si
PH100-SiUV
PH20-Ge
PRONTO-Si
UM-B
XLP12-HpW
12nW

gentec-eo.com/laser-power-meter

µW

mW

W

kW

MW

POWER DETECTORS
General use power detectors

Available with

UP-H: BROADBAND THERMAL DETECTORS• Available in 6 sizes:

10 mm Ø12 mm Ø17 mm Ø19 mm Ø25 mm Ø55 mm Ø• Available with 5 cooling modules:

ENERGY DETECTORS

Convection (S)Small heatsink (H)Large heatsink (L)Fan (F)
Water (W)

THE WIDEST RANGE OF LASER MEASUREMENTS

ew

BEAM PROFILING

PRONTO-250-FLEX

t

c
du

o
pr

• Compact laser power meter up to 250 W

N• Three measurement modes :TERAHERTZ DETECTORSSingle-Shot Power (SSP): up to 250 W
Continuous Power (CWP): up to 8 W
Single Shot Energy (SSE): up to 25 J
• The FLEXibility to pick only the calibrations you need:

Default calibration "Y": for visible to NIR wavelengths (248 nm to 2.5 µm)
Additional calibration "C": for CO2 lasers (10.6 µm)
Additional calibration "E": for energy measurements with ± 5 % accuracy

PORTABLE & EASY TO USEDISPLAYS & PC INTERFACESCOMPARISON TABLE - GENERAL USE POWER DETECTORS

pW

nW

µW

mW
T 418.651.8003 | info@gentec-eo.com

W

kW

CUSTOM / OEM PRODUCTS

Spectral range
0.19 - 20 µm
0.19 - 20 µm
0.19 - 20 µm
0.19 - 20 µm
0.19 - 20 µm
0.19 - 20 µm
0.19 - 20 µm
0.19 - 20 µm

UP10-H
UP12-H
UP17-H
UP19-HPRONTO-250
UP25-H
UP55-H
UP55-HDPRODUCT GUIDE 2024

POWER DETECTORS

Our standard absorber offers high damage thresholds and a flat spectral
response, making this series of power detectors a versatile solution that can covermost of your laser power measurement needs.

MW
.com

13

POWER DETECTORSHigh performance power detectors

Available with

UP-W
Our “W” absorber can handle tightly focused beams thanks to its extremely highdamage threshold for average power density. It can be used to measure up to 50
W, from the UV to IR.
• Available in 2 sizes: 19 mm or 50 mm Ø aperture• High damage threshold absorber (100 kW/cm2)• Our highest maximum average power density

IDEAL FOR UV LASERS & TIGHTLY FOCUSED BEAMS
Available with

UP-QEDThe UP-QED series are power detectors for lasers with extreme power and energy
density, such as laser micromachining systems. Thanks to a proprietary absorberthat diffuses the measured beam and absorbs it in a larger volume, thesedetectors have the highest damage thresholds on the market.
• Available in 2 sizes: 16 mm or 52 mm Ø aperture• Our highest maximum average power density• Our highest maximum energy density• Not suitable for UV lasers

THE HIGHEST DAMAGE THRESHOLDS ON THE MARKET!
Available with

IS: INTEGRATING SPHERE POWER METER
Get the best of both worlds with our new integrating sphere power meters. This
technology offers the fast risetime of photodetectors with the high average powerof thermal detectors.
• Fast risetime: 0-95 % in less than 0.2 seconds• Measures up to 1000 W of continuous power• Available in 2 sizes: 12 mm or 50 mm Ø aperture• Integrated signal processing with USB or RS-232 output

FAST AND ROBUST POWER MEASUREMENTCOMPARISON TABLE - HIGH PERFORMANCE POWER DETECTORS
Spectral range
0.19 - 10 µm
0.19 - 10 µm
0.19 - 10 µm
0.27 - 2.5 µm
0.27 - 2.5 µm
0.34 - 1.1 µm
0.34 - 1.1 µm

UP17-W
UP19-W
UP50-W
UP16-QED
UP52-QED
IS12
IS50
pW
14

nW

gentec-eo.com/laser-power-meter

µW

mW

W

kW

MW

POWER DETECTORSHigh power measurement

Available with

HP60: HIGH POWER, LOW BACK-REFLECTIONSFOR SMALL BEAMS UP TO 15 KW

POWER DETECTORSThe gold reflector cone of the HP60 series is specifically designed to handle the highintensities of very small beams. By reflecting the incident light on the sides of theaperture, the cone effectively spreads the intensity on a larger area, thus raising thedamage threshold to 10 kW/cm2 at the full power (15 kW).

Available withThe HP100A and HP125A are the smallest in our HP series of high-power detectors.
They are versatile high-power detectors that measure up to 15 kW of continuous
power with a noise level of only a few watts. These models feature a very large
aperture of 100 or 125 mm Ø.

CONTINUOUS POWER MEASUREMENT UP TO 15 KW

ENERGY DETECTORS

HP100/125: LARGE APERTURE, COMPACT DEVICE

Available with

Our unique high-power design allows for infinite customization capabilities. Do nothesitate to contact us with your specific needs. Our Super HP models feature a USB
output for direct measurements on a PC as well as our standard DB15 connector.
Wireless output is also available.

BEAM PROFILING

SUPER HP: CUSTOM, HIGH-POWER MEASUREMENTCUSTOM SOLUTIONS FOR UP TO 150 KWWhen you are on the go and water cooling is not easily accessible, the PRONTO
high-power probes are the best solution. These all-in-one power meters withtouchscreen controls come in 4 models: 500 W, 3 kW, 6 kW and 10 kW. Theirintegrated display is encased in a rugged metallic casing to withstand the harshestof environments.

DISPLAYS & PC INTERFACESUP TO 10 KW WITHOUT WATER COOLINGCOMPARISON TABLE - HIGH POWER MEASUREMENTPRODUCT GUIDE 2024

nW

µW

mW

T 418.651.8003 | info@gentec-eo.com

W

kW

CUSTOM / OEM PRODUCTS

Spectral range
0.80 - 12 µm
0.19 - 20 µm
0.19 - 20 µm
0.19 - 20 µm
0.19 - 20 µm

HP60
HP100
HP125
SUPER HP
PRONTO (high power)pW

TERAHERTZ DETECTORSPRONTO: PORTABLE HIGH-POWER PROBES

MW
.com

15

PH

10 pW to 750 mW, Si and Ge sensors

OD ATTENUATORSOD attenuators sold inoption. When bought
together, the detector iscalibrated with and without
the attenuator.

PH series detector
with OD attenuator

OUTPUT OPTIONSSMART DB15 CONNECTOR
Contains all the calibration data
ALL-IN-ONE-METER
Connects directly to a PC
Two models available: 				USB output (-INT)
RS-232 output (-IDR)

COMPATIBLE DISPLAYS & PC INTERFACESKEY FEATURESMIRO ALTITUDE

MAESTRO

TUNER

UNO

U-LINK and P-LINK

M-LINK

Stand with delrin post

Extension cables(4, 15, 20 or 25 m)

Fiber adaptors & connectors(FC, SC, ST and SMA)OD attenuators

Pelican carrying case

Isolation tube

LARGE APERTURES
10 mm Ø for the silicon sensors
3 VERSIONS• Silicon: 350 - 1080 nm, up to 750 mW• Silicon-UV: 210 - 1080 nm, up to 38 mW• Germanium: 800 - 1650 nm, up to 500 mW
CHOICE OF ATTENUATORSACCESSORIES• OD0.3: 50 % transmission (for PH100-SIUV only)• OD1: 10 % transmission• OD2: 1 % transmission

HIGH ACCURACY
The PH100-SI-HA presents the lowest calibration
uncertainty to date
PRECISE CALIBRATION
Wavelength selection in 1 nm steps

16

gentec-eo.com/laser-power-meter

PHSpecifications

PH100-SIUV-D0

PH20-GE-D0(ALONE / WITH MAX ATTENUATION)

36 mW / 750 mW

4 mW / 38 mW

30 mW / 500 mW

EFFECTIVE APERTURE

10 mm Ø10 mm Ø5 mm Ø

350 - 1080 nm

210 - 1080 nm

800 - 1650 nm

With OD0.3---

210 - 1080 nm---

With OD1

400 - 1080 nm

400 - 1080 nm

900 - 1650 nm

With OD2

630 - 1080 nm---

950 - 1650 nm

MAX AVERAGE POWER*

POWER DETECTORS

PH100-SI-HA-D0

MEASUREMENT CAPABILITY
Calibrated spectral range

36 mW at 1064 nm

4 mW at 532 nm

30 mW at 1064 nm

With OD0.3---

16 mW at 300 nm---

With OD1

300 mW at 1064 nm

38 mW at 532 nm

300 mW at 1064 nm

With OD2

750 mW at 1064 nm---

500 mW at 1064 nm

Noise equivalent power a

10 pW at 980 nm

10 pW at 850 nm

60 pW at 1550 nm

Rise time (nominal)

0.2 s

0.2 s

0.2 s

± 5.0  % (350 - 399 nm)± 18  % (210 - 229 nm)± 5.0  % (800 - 1049 nm)± 2.0  % (400 - 449 nm)± 8.0  % (230 - 254 nm)± 3.5 % (1050 - 1559 nm)± 1.5 % (450 - 809 nm)± 6.5 % (255 - 399 nm)± 7.0 % (1560 - 1629 nm)± 2.5 % (400 - 899 nm)± 10 % (1630 - 1650 nm)± 4.0 % (900 - 1009 nm)---± 7.5 % (1010 - 1080 nm)± 7.5 % (1010 - 1080 nm)---± 5.0 % (400 - 419 nm)

Same as without attenuator± 5.0 % (900 - 1559 nm)± 4.0 % (420 - 899 nm)---± 7.0 % (1560 - 1629 nm)± 5.0 % (900 - 1009 nm)---± 10 % (1630 - 1650 nm)± 7.5 % (1010 - 1080 nm)

---

---

155 kHz155 kHz155 kHz

100 W/cm2

100 W/cm2

100 W/cm2

Effective aperture

10 mm Ø10 mm Ø5 mm Ø

Distance to sensor face

13.7 mm

13.7 mm10.5 mm

SensorSiliconUV-Silicon

Germanium

Dimensions

38.1Ø x 27.4D mm

38.1Ø x 27.4D mm

38.1Ø x 27.4D mm

Weight (head only)

130 g

130 g

130 g

DB15, USB or RS-232

DB15, USB or RS-232

DB15, USB or RS-232

STAND-D-233 or STAND-D-233-M

STAND-D-233 or STAND-D-233-M

STAND-D-233 or STAND-D-233-M

Calibration uncertainty (with OD filters)

Minimum repetition rate

TERAHERTZ DETECTORS± 2.0 % (810 - 899 nm)± 4.0 % (900 - 1009 nm)

BEAM PROFILING

Calibration uncertainty

ENERGY DETECTORS

Maximum measurable power*

DAMAGE THRESHOLDS
Maximum average power density
PHYSICAL CHARACTERISTICS

Compatible stand

DISPLAYS & PC INTERFACESORDERING INFORMATION
Available output options

CUSTOM / OEM PRODUCTS

Product page

* See curves (p. 62-64) for maximum power at other wavelengths
a. Nominal value. Depends on environmental electromagnetic interference and wavelength.

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

17

PRONTO-SI

0.3 nW - 800 mW power probe with touchscreen controls

Flips closed
to protect the display and sensor

Silicon sensor

USER INTERFACE

Slide-in attenuator

3 Displays for the measurements
Real-time display

Bargraph display

Min/Max display

Displays the measured
value with large digits
so you can see themfrom a distance

Adds a bargraph below
the measured value, for
an intuitive understandingof the trend of your laser

In addition to the realtime value, the device
displays the lowest
and highest values

Save your data andtransfer it to your PC

Adjust the
wavelength

Set the brightness
and orientation

Slim profile

8-32 mounting
hole for post

USB connector for:- Battery charging- Data transfer to PC- Internet upgrades

Color LCD display withtouch screen controls
On/Offand main menu

KEY FEATURES
	POCKET-SIZE
This low power laser probe is so compact it fits in yourpocket!
SLIM PROFILE
The sensor part is only 6 mm thick, allowing it to fit
into tight spaces
EASY TO USE
The color LCD touchscreen allows for a friendly userinterface. You can make a measurement with just thetouch of a button!

DATA TRANSFER TO PC

VERY LOW POWER MEASUREMENTS
Thanks to its very low noise level of only 10 pW, the
PRONTO-Si measures powers as low as 0.3 nW
SLIDE-IN ATTENUATOR
Just slide the OD1 integrated filter to the ON positionand you can measure up to 800 mW of continuous
power at 532 nm (maximum power varies withwavelength)
DATA LOGGING
Save your data to the internal memory and thentransfer them to your PC over the USB connection

ACCESSORIESOPTIONAL FIBER OPTICS ADAPTOR
The fiber optics adaptor is held securely in place witha set screw and is compatible with OD attenuators
SERIAL COMMANDS
Serial commands are available to let you take fullcontrol of your PRONTO from your PC.

18

gentec-eo.com/laser-power-meter

Threaded adaptor
for PRONTO-Si

Fiber adaptors

PRONTO-SI
Specifications

POWER DETECTORSPRONTO-SI
MAX AVERAGE POWER*(ATTENUATOR OFF / ATTENUATOR ON)

88 mW / 800 mW

EFFECTIVE APERTURE

10 x 10 mm

INTERFACE

Touchscreen color LCD display

Attenuator OFF

320 - 1100 nm

Attenuator ON

400 - 1100 nm

ENERGY DETECTORSMEASUREMENT CAPABILITY
Calibrated spectral range

Power range*
Attenuator OFF

0.3 nW - 88 mW at 532 nm

Attenuator ON

3 nW - 800 mW at 532 nm
10 pW at 980 nm

Response time

0.2 s

Measurement accuracy

From ± 1.5 % to ± 7.5 % (wavelength-dependent )Display resolution

1 pW

DAMAGE THRESHOLDS
Maximum average power density
Maximum average power
USER INTERFACE
Displays

100 W/cm2
800 mW (with attenuator ON)

Real-time, bar graph and min/max
Zero offset, wavelength selection and reset data

Data acquisition and transfer

Yes

Touchscreen Color LCD

Display size

28.0 x 35.0 mm (128 x 160 pixels)

Data storage

50 000 pts

Battery type

Rechargeable Li-ion

Battery life

17 hours (with brightness set at 25 %)Battery recharge via

USB port

10 x 10 mm

SensorSiliconAttenuator

Integrated slide-in OD1 attenuator

Mounting hole (for post)

1 x 8-32

Dimensions (Open)41W x 216.2L x 15.8D mm (Sensor part is only 6.0D mm)

Dimensions (Closed)

41W x 136L x 22.1D mm

Weight

150 g
CUSTOM / OEM PRODUCTS

ORDERING INFORMATION
Compatible stand

DISPLAYS & PC INTERFACESPHYSICAL CHARACTERISTICS
Effective aperture

TERAHERTZ DETECTORS

Measurement controls
GENERAL SPECIFICATIONS
Display type

BEAM PROFILING

Noise equivalent power

STAND-S-233

Product page

* See curves (page 65) for maximum power at other wavelengths
PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

19

UM-B

5 nW - 25 mW, radiometer for utra-low power measurements

OUTPUT OPTIONSSMART DB15 CONNECTOR
Contains all the calibration data
Included in UM9B-BL-D0 model only
ANALOG OUTPUT
When used with APM (D) analog power supply

COMPATIBLE DISPLAYS & PC INTERFACES

U-LINKAPM (D) analog power supply(for UM9B-BL-L-D0)

Stand with delrin post

SDC-500 digital
optical chopper

Extra isolation tube

Fiber adaptors & connectors(FC, ST and SMA)

Pelican carrying case

MAESTROKEY FEATURESACCESSORIESVERY LOW NOISE LEVEL
Noise levels of a photodetector, but with the large
bandwidth of a pyroelectric:• Down to 5 nW when using the analog power
module (APM)
VERY HIGH RESPONSIVITY
Up to 20 000 V/W when using the analog power
module (APM)
VERY LARGE BANDWIDTH
From DUV to FIR thanks to pyroelectric technology
	INCLUDES AN ISOLATING TUBE TO
BLOCK UNDESIRED NOISE FROM THE
ENVIRONMENT.

20

gentec-eo.com/laser-power-meter

UM-B
Specifications

UM9B-BL-D0

MAX AVERAGE POWER

200 μW

20 mW (MAESTRO), 25 mW (U-LINK)EFFECTIVE APERTURE

9 mm Ø

9 mm Ø

COMPATIBLE DISPLAYS & PC INTERFACES

APM (D)

MAESTRO and U-LINK

POWER DETECTORS

UM9B-BL-L-D0

MEASUREMENT CAPABILITY
0.1 - 20 μm

0.1 - 20 μm

Calibrated spectral range

633 nmb

0.248 - 2.1 µma

Maximum measurable power

200 μW

20 mW (MAESTRO), 25 mW (U-LINK)

Noise equivalent power (RMS)

5 nW

300 nW

Rise time (0-100 %)≤ 0.2 s≤ 0.2 sCalibration uncertainty± 4 % at 1064 nm± 4 % at 1064 nm

Chopper frequency

5 ± 1 Hz10 ± 1 Hz

50 mW/cm2

50 mW/cm2

Effective aperture

9 mm Ø

9 mm Ø

SensorPyroelectric

Pyroelectric

Absorber

BL

BL

Dimensions

38.1Ø X 79D mm

38.1Ø X 79D mm

Weight91 g

91 g

DB15 only

DB15 only

STAND-D-233 or STAND-D-233-M

STAND-D-233 or STAND-D-233-M

ENERGY DETECTORS

Spectral range

DAMAGE THRESHOLDS
Maximum average power density (1064 nm)

Compatible stand

TERAHERTZ DETECTORSORDERING INFORMATION
Available output options

BEAM PROFILINGPHYSICAL CHARACTERISTICS

Product pageDISPLAYS & PC INTERFACES

a. Calibrations at 2.1 to 2.5 µm and 10.6 µm are available on special request.
b. Typical wavelength correction factors are provided for 0.19 to 2.1 µm.

CUSTOM / OEM PRODUCTS

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

21

XLP1212 mm Ø, 0.5 µW - 3 W, low power thermopile

OUTPUT OPTIONSSMART DB15 CONNECTOR
Contains all the calibration data
ALL-IN-ONE-METER
Connects directly to a PC
Two models available: 				USB output (-INT)
RS-232 output (-IDR)

COMPATIBLE DISPLAYS & PC INTERFACESKEY FEATURESMIRO ALTITUDE

MAESTRO

TUNER

UNO

U-LINK and P-LINK

S-LINK and M-LINK

Stand with steel post

Extension cables(4, 15, 20 or 25 m)

IR filter(Mounted)

Fiber adaptors & connectors(FC, ST and SMA)

Pelican carrying case

Extra isolation tube

ACCESSORIESLOW POWER THERMOPILE
Noise level of a photodetector with the large bandwidthand high power capacity of a thermal device
MINIMAL THERMAL DRIFT
Only 6 µW/°C (with the IR filter)
HIGH SENSITIVITY
SPECIAL MODEL FOR ULTRASHORT PULSES
VP (volume absorber) version is perfect for low power
lasers with ultrashort pulses (ps and fs)
IR FILTER (XLPF12 MODEL)
Removes unwanted IR interference
ISOLATION TUBE
Eliminates power fluctuations created by air turbulence

22

gentec-eo.com/laser-power-meter

XLP12
Specifications

XLP12-3S-VP-D0(CONTINUOUS / 1 MINUTE)

3W/3W
Broadband absorber, with IR filter

3W/3W
Volume absorber

EFFECTIVE APERTURE

12 mm Ø12 mm Ø12 mm ØCOOLING METHOD

Convection

Convection

Convection

MEASUREMENT CAPABILITY
Spectral range

0.19 - 20 μm

0.28 - 2.1 μm

0.248 - 20 μm

Calibrated spectral range a

0.248 - 2.1 µm

0.308 - 2.1 µm

0.248 - 2.1 µm

Noise equivalent power b

0.5 µW

0.5 µW

0.5 µW

Thermal drift c

12 µW/°C

6 µW/°C

12 µW/°C

Rise time (nominal)

2.5 s

2.5 s

3s

Calibration uncertainty e± 2.5 %

± 2.5 %

± 2.5 %

Repeatability±0.5 %±0.5 %±0.5 %

Maximum measurable energy f

5J

5J

---

Noise equivalent energyd

ENERGY DETECTORSXLPF12-3S-H2-D0

3W/3W
Broadband absorber

POWER DETECTORSXLP12-3S-H2-D0
MAX AVERAGE POWER

Energy mode
---

16 s

---

Maximum pulse width

300 ms

300 ms

---

± 5%

---

1 kW/cm2

1 kW/cm2

30 W/cm2 at 1064 nm
8 W/cm2 at 532 nm
4 W/cm2 at 355 nm

1064 nm, 360 µs, 5 Hz

5 J/cm2

5 J/cm2---

1064 nm, 7 ns, 10 Hz

1 J/cm

1 J/cm2

4 J/cm2

532 nm, 7 ns, 10 Hz

0.6 J/cm2

0.6 J/cm2

3 J/cm2

355 nm, 7 ns, 10 Hz

---

---

1 J/cm2

266 nm, 7 ns, 10 Hz

0.3 J/cm

0.3 J/cm---

PHYSICAL CHARACTERISTICS
Effective aperture

12 mm Ø12 mm Ø12 mm ØAbsorber (high damage threshold)

H2

H2

VP (Volume absorber)

Dimensions

73H x 73W x 20D mm(72D mm with tube)

73H x 73W x 20D mm(80D mm with tube)

73H x 73W x 20D mm(72D mm with tube)

Weight (head only)

0.31 kg

0.32 kg

0.32 kg

DB15, USB, RS-232 or Bluetooth

DB15, USB, RS-232 or Bluetooth

DB15, USB, RS-232 or Bluetooth

STAND-S-233

STAND-S-233

STAND-S-233

Accuracy with energy calibration option ± 5 %DAMAGE THRESHOLDS
Maximum average power density g

TERAHERTZ DETECTORS

12 µJ

16 s

BEAM PROFILING

12 µJ

Minimum repetition periodbMaximum energy density

Compatible stand

2

2

DISPLAYS & PC INTERFACESORDERING INFORMATION
Available output options

2

Product page

PRODUCT GUIDE 2024

T 418.651.8003 | info@gentec-eo.com

CUSTOM / OEM PRODUCTS

a. Calibrations at 2.1 to 2.5 µm and 10.6 µm are available on special request.
b. Nominal value, actual value depends on electrical noise in the measurement system.
c. With Gentec-EO MAESTRO.
d. With anticipation.
e. Including linearity with power.
f. For 360 µs pulses. Higher pulse energy possible for long pulses (ms), less for short pulses (ns).
g. At 1064 nm, 1 W CW.
Specifications are subject to change without notice

.com

23

UP10-H

10 mm Ø, 0.1 mW - 2 W, fast & low power thermopile

OUTPUT OPTIONSSMART DB15 CONNECTOR
Contains all the calibration data
ALL-IN-ONE-METER
Connects directly to a PC
Two models available: 				USB output (-INT)
RS-232 output (-IDR)

COMPATIBLE DISPLAYS & PC INTERFACESKEY FEATURESMIRO ALTITUDE

MAESTRO

TUNER

UNO

U-LINK and P-LINK

S-LINK and M-LINK

Stand with steel post

Extension Cables(4, 15, 20 or 25 m)

IR Filter(Mounted)

Isolation tube

Fiber adaptors & connectors(FC, ST and SMA)

Pelican carrying case

ACCESSORIESLOW POWER THERMOPILE
Noise level of a photodetector with thelarge bandwidth and high power capacityof a thermal device
HIGH PERFORMANCE
Fast rise time (1.4 s)High damage threshold (36 kW/cm2)
COMPACT DESIGN
Only 13 mm thick (UP10P model)
ENERGY MODE
Measure single shot energy up to 3 J

24gentec-eo.com/laser-power-meterUP10-H
Specifications

UP10K-2S-H5-L-D0

MAX AVERAGE POWER

2W

2W

EFFECTIVE APERTURE

10 mm Ø10 mm ØCOOLING METHOD

Convection

Convection

Spectral range

0.19 - 20 µm

0.19 - 20 µm

Calibrated spectral range a

0.248 - 2.1 µm

0.248 - 2.1 µm

Noise equivalent power b

100 μW without anticipation
30 μW with anticipation and 2 s moving average

100 μW without anticipation
30 μW with anticipation and 2 s moving average

Rise time (nominal) c

1.4 s

1.1 s

Calibration uncertainty d± 2.5 %

± 2.5 %

Repeatability±0.5 %±0.5 %

Maximum measurable energy e

3J

3J

Noise equivalent energy b

5 mJ

5 mJ

Minimum repetition period

2s

2s

Maximum pulse width

63 ms

63 ms

Accuracy with energy calibration option

± 5%

± 5%

36 kW/cm2

36 kW/cm2

1064 nm, 360 µs, 5 Hz

5 J/cm2

5 J/cm2

1064 nm, 7 ns, 10 Hz

1 J/cm

1 J/cm2

532 nm, 7 ns, 10 Hz

0.6 J/cm2

0.6 J/cm2

266 nm, 7 ns, 10 Hz

2

0.3 J/cm

0.3 J/cm2

Effective aperture

10 mm Ø10 mm ØAbsorber (high damage threshold)

H5

H5

Dimensions

46H x 46W x 13D mm

50H x 50W x 21.5D mm

Weight (head only)

0.13 kg

0.19 kg

Available output options

DB15, USB or RS-232

DB15, USB, RS-232

Compatible stand

STAND-S-233

STAND-S-233

POWER DETECTORSUP10P-2S-H5-L-D0

MEASUREMENT CAPABILITY
ENERGY DETECTORS

Energy mode

BEAM PROFILINGDAMAGE THRESHOLDS
Maximum average power density f
Maximum energy density

TERAHERTZ DETECTORS

2

PHYSICAL CHARACTERISTICSDISPLAYS & PC INTERFACESORDERING INFORMATION

Product page

CUSTOM / OEM PRODUCTS

a. Calibrations at 2.1 to 2.5 µm and 10.6 µm are available on special request.
b. Nominal value, actual value depends on electrical noise in the measurement system.
c. With anticipation.
d. Including linearity with power.
e. For 360 µs pulses. Higher pulse energy possible for long pulses (ms), less for short pulses (ns).
f. At 1064 nm, 10 W CW.

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

25

UP12-H

12 mm Ø, 1 mW - 110 W

OUTPUT OPTIONSSMART DB15 CONNECTOR
Contains all the calibration data
ALL-IN-ONE-METER
Connects directly to a PC
Two models available: 				USB output (-INT)
RS-232 output (-IDR)

COMPATIBLE DISPLAYS & PC INTERFACESKEY FEATURESMIRO ALTITUDE

MAESTRO

TUNER

UNO

U-LINK and P-LINK

S-LINK and M-LINK

Extension cables(4, 15, 20 or 25 m)

Replacement coverfor fiber adaptors

ACCESSORIESMODULAR CONCEPT
Increase the power capability of yourdetector: 3 different cooling modules
HIGH PERFORMANCE
Fast rise time (0.3 s)High damage threshold (36 kW/cm2)
COMPACT DESIGN
Only 14 mm thick (10S model)Stand with steel post

ENERGY MODE
Measure single shot energy up to 5 J

Pelican carrying Case

26

gentec-eo.com/laser-power-meter

UP12-H
Specifications

UP12E-20H-H5-D0

UP12E-70W-H5-D0

10 W / 20 W

20 W / 40 W

70 W f / 110 W f

EFFECTIVE APERTURE

12 mm Ø12 mm Ø12 mm ØCOOLING METHOD

Convection

Heatsink

Water-cooledSpectral range

0.19 - 20 μm

0.19 - 20 μm

0.19 - 20 μm

Calibrated spectral range a

0.248 - 2.1 µm

0.248 - 2.1 µm

0.248 - 2.1 µm

Noise equivalent power b

1 mW

1 mW

1 mW

Rise time (nominal)(CONTINUOUS / 1 MINUTE)

POWER DETECTORSUP12E-10S-H5-D0
MAX AVERAGE POWER

MEASUREMENT CAPABILITY

0.3 s

0.3 s

± 2.5 %

± 2.5 %

± 2.5 %

Repeatability±0.5 %±0.5 %±0.5 %

Maximum measurable energy e

5J

5J

5J

Noise equivalent energy b

0.02 J

0.02 J

0.02 J

Minimum repetition period

1.5 s

1.5 s

1.5 s

Maximum pulse width

50 ms

50 ms

50 ms

Accuracy with energy calibration option

± 5%

± 5%

± 5%

36 kW/cm2

36 kW/cm2

36 kW/cm2

1064 nm, 360 µs, 5 Hz

5 J/cm2

5 J/cm2

5 J/cm2

1064 nm, 7 ns, 10 Hz

1 J/cm

1 J/cm

1 J/cm2

532 nm, 7 ns, 10 Hz

0.6 J/cm2

0.6 J/cm2

0.6 J/cm2

266 nm, 7 ns, 10 Hz

0.3 J/cm

0.3 J/cm

0.3 J/cm2

Effective aperture

12 mm Ø12 mm Ø12 mm ØAbsorber (high damage threshold)

H5

H5

H5

Dimensions

38H x 38W x 14D mm

38H x 38W x 45D mm

38H x 38W x 32D mm

Weight (head only)

0.13 kg

0.15 kg

0.19 kg

Available output options

DB15, USB or RS-232

DB15, USB or RS-232

DB15, USB or RS-232

Compatible stand

STAND-S-233

STAND-S-233

STAND-S-233

ENERGY DETECTORS

0.3 s

Calibration uncertainty dcEnergy mode

BEAM PROFILINGDAMAGE THRESHOLDS
Maximum average power density g
Maximum energy density22

2

TERAHERTZ DETECTORS

2

PHYSICAL CHARACTERISTICSDISPLAYS & PC INTERFACESORDERING INFORMATION

Product page

CUSTOM / OEM PRODUCTS

a. Calibrations at 2.1 to 2.5 µm and 10.6 µm are available on special request.
b. Nominal value, actual value depends on electrical noise in the measurement system.
c. With anticipation.
d. Including linearity with power.
e. For 360 μs pulses. Higher pulse energy possible for long pulses (ms), less for short pulses (ns).
f. Minimum cooling flow 0.5 liters/min, water temperature ≤ 22°C, 1/8 NPT compression fittings for 1/4 inch semi-rigid tube.
Contact Gentec-EO for clean deionized water cooling module option.
g. At 1064 nm, 10 W CW.

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

27

UP17-H/W

17 mm Ø, 1 mW - 7 W, ultra thin casing

OUTPUT OPTIONSSMART DB15 CONNECTOR
Contains all the calibration data
ALL-IN-ONE-METER
Connects directly to a PC
Two models available: 				USB output (-INT)
RS-232 output (-IDR)

COMPATIBLE DISPLAYS & PC INTERFACESKEY FEATURESMIRO ALTITUDE

MAESTRO

TUNER

UNO

U-LINK and P-LINK

S-LINK and M-LINK

Extension cables(4, 15, 20 or 25 m)

Pelican carrying case

ACCESSORIESULTRA THIN CASING
Only 10.7 mm thick!
CHOICE BETWEEN 2 ABSORBERS• H5: 36 kW/cm2• W5: unequalled 100 kW/cm2
HIGH POWER TO SIZE RATIO
6 W continuous reading
ENERGY MODE
Measure single shot energy up to 200 J(with the W5 version)

28

gentec-eo.com/laser-power-meterStand with steel post

UP17-H/W
Specifications

UP17P-6S-W5-D0

6W/7W

6W/7W

EFFECTIVE APERTURE

17 mm Ø17 mm ØCOOLING METHOD

Convection

Convection

Spectral range

0.19 - 20 μm

0.19 - 10.0 μm

Calibrated spectral range

0.248 - 2.1 µm a

0.248 - 2.1 µm b

Noise equivalent power c

1 mW

1 mW

Rise time (nominal)(CONTINUOUS / 1 MINUTE)

POWER DETECTORSUP17P-6S-H5-D0
MAX AVERAGE POWER

MEASUREMENT CAPABILITY

1.4 s± 2.5 %

± 2.5 %

Repeatability±0.5 %±0.5 %

Maximum measurable energy f

15 J

200 J

Noise equivalent energy c

0.02 J

0.02 J

Minimum repetition period

4s

5s

Maximum pulse width

88 ms

133 msENERGY DETECTORS

0.8 s

Calibration uncertainty edEnergy mode

BEAM PROFILING	Accuracy with energy calibration option ± 5 %± 5%

DAMAGE THRESHOLDS
Maximum average power density g

36 kW/cm2

100 kW/cm2

1064 nm, 360 µs, 5 Hz

5 J/cm2

100 J/cm2

1064 nm, 7 ns, 10 Hz

1 J/cm

1.1 J/cm2

532 nm, 7 ns, 10 Hz

0.6 J/cm2

1.1 J/cm2

266 nm, 7 ns, 10 Hz

0.3 J/cm

0.7 J/cm2

Effective aperture

17 mm Ø17 mm ØAbsorber (high damage threshold)

H5W5

Dimensions

46H x 46W x 10.7D mm

46H x 46W x 10.7D mm

Weight (head only)

0.1 kg

0.1 kgAvailable output options

DB15, USB or RS-232

DB15, USB or RS-232

Compatible stand

STAND-S-233

STAND-S-233

Maximum energy density

2

TERAHERTZ DETECTORS

2

PHYSICAL CHARACTERISTICSDISPLAYS & PC INTERFACESORDERING INFORMATION

Product page

CUSTOM / OEM PRODUCTS

a. Calibrations at 2.1 to 2.5 µm and 10.6 µm are available on special request.
b. Calibration at 2.1 to 2.5 µm is available on special request.
c. Nominal value, actual value depends on electrical noise in the measurement system.
d. With anticipation.
e. Including linearity with power.
f. For 360 µs pulses. Higher pulse energy possible for long pulses (ms), less for short pulses (ns).
g. At 1064 nm, 10 W CW.

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

29

UP19-H

19 mm Ø, 1 mW - 200 W

OUTPUT OPTIONSSMART DB15 CONNECTOR
Contains all the calibration data
ALL-IN-ONE-METER
Connects directly to a PC
Two models available: 				USB output (-INT)
RS-232 output (-IDR)
BLU WIRELESS METER
Connects via Bluetooth® to a smartphone, tablet or PC

COMPATIBLE DISPLAYS & PC INTERFACESKEY FEATURESMIRO ALTITUDE

MAESTRO

TUNER

UNO

U-LINK and P-LINK

S-LINK and M-LINK

Stand with steel post

Extension cables(4, 15, 20 or 25 m)

Isolation tube

Fiber adaptors and connectors(FC, SC or SMA)

12V power supply

Pelican carrying case

ACCESSORIESMODULAR CONCEPT
Increase the power capability of yourdetector: 5 different cooling modules
HIGH PERFORMANCE
Fast Rise Time (0.6 s)High damage threshold (45 kW/cm2)
COMPACT DESIGN
Only 20.6 mm thick (15S model)
ENERGY MODE
Measure single shot energy up to 25 J

30gentec-eo.com/laser-power-meter

UP19-H
Specifications

UP19K-30H-H5-D0 UP19K-50L-H5-D0 UP19K-110F-H9-D0 UP19K-200W-H9-D0

15 W / 30 W

30 W/ 60 W

50 W / 90 W

110 W / 150 W

200 W f / 200 W f

EFFECTIVE APERTURE

19 mm Ø19 mm Ø19 mm Ø19 mm Ø19 mm ØCOOLING METHOD

Convection

Heatsink

Large heatsink

Fan-cooled

Water-cooledSpectral range

0.19 - 20 μm

0.19 - 20 μm

0.19 - 20 μm

0.19 - 20 μm

0.19 - 20 μm

Calibrated spectral range a

0.248 - 2.1 µm

0.248 - 2.1 µm

0.248 - 2.1 µm

0.248 - 2.1 µm

0.248 - 2.1 µm

Noise equivalent power b

1 mW

1 mW

1 mW

3 mW

3 mW

Rise time (nominal)(CONTINUOUS / 1 MINUTE)

POWER DETECTORSUP19K-15S-H5-D0
MAX AVERAGE POWER

MEASUREMENT CAPABILITY

0.6 s

0.6 s

1.5 s

1.5 s

± 2.5 %

± 2.5 %

± 2.5 %

± 2.5 %

± 2.5 %

Repeatability±0.5 %±0.5 %±0.5 %±0.5 %±0.5 %

Maximum measurable energy e

15 J

15 J

15 J

25 J

25 J

Noise equivalent energy b

0.02 J

0.02 J

0.02 J

0.06 J

0.06 J

Minimum repetition period

4s

4s

4s

4s

4s

Maximum pulse width

88 ms

88 ms

88 ms

88 ms

88 ms

Accuracy with energy calibration option ± 5 %

± 5%

± 5%

± 5%

± 5%

36 kW/cm2

36 kW/cm2

36 kW/cm2

45 kW/cm2

45 kW/cm2

1064 nm, 360 µs, 5 Hz

5 J/cm2

5 J/cm2

5 J/cm2

5 J/cm2

5 J/cm2

1064 nm, 7 ns, 10 Hz

1 J/cm

1 J/cm

1 J/cm

1 J/cm

1 J/cm2

532 nm, 7 ns, 10 Hz

0.6 J/cm2

0.6 J/cm2

0.6 J/cm2

0.6 J/cm2

0.6 J/cm2

266 nm, 7 ns, 10 Hz

0.3 J/cm

0.3 J/cm

0.3 J/cm

0.3 J/cm

0.3 J/cm2

Effective aperture

19 mm Ø19 mm Ø19 mm Ø19 mm Ø19 mm ØAbsorber (high damage threshold)

H5

H5

H5

H9

H9

Dimensions

50H x 50W
x 20.6D mm

50H x 50W
x 56.3D mm

76.2H x 76.2W
x 73.6D mm

50H x 50W
x 63D mm

50H x 50W
x 33D mm

Weight (head only)

0.16 kg

0.21 kg

0.48 kg

0.25 kg

0.24 kg

Available output options

DB15, USB, RS-232 orBluetooth

DB15, USB, RS-232 orBluetooth

DB15, USB or RS-232

DB15, USB, RS-232 orBluetooth

DB15, USB, RS-232 orBluetoothCompatible stand

STAND-S-233

STAND-S-233

STAND-S-233

STAND-S-233

STAND-S-233

ENERGY DETECTORS

0.6 s

Calibration uncertainty dcEnergy mode

BEAM PROFILINGDAMAGE THRESHOLDS
Maximum average power density g
Maximum energy density
2

2

2

2

2

2

2

TERAHERTZ DETECTORS

2

PHYSICAL CHARACTERISTICSDISPLAYS & PC INTERFACESORDERING INFORMATION

Product page

CUSTOM / OEM PRODUCTS

a. Calibrations at 2.1 to 2.5 µm and 10.6 µm are available on special request.
b. Nominal value, actual value depends on electrical noise in the measurement system.
c. With anticipation.
d. Including linearity with power.
e. For 360 μs pulses. Higher pulse energy possible for long pulses (ms), less for short pulses (ns).
f. Minimum cooling flow 0.5 liters/min, water temperature ≤ 22°C, 1/8 NPT compression fittings for 1/4 inch semi-rigid tube.
Contact Gentec-EO for clean deionized water cooling module option.
g. At 1064 nm, 10 W CW.
PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

31

PRONTO

0.5 W - 250 W power probes with touchscreen controls

Patterned surface forenhanced cooling Rubber band for a betterstability on a flat surface
High damage thresholdbroadband absorber

USER INTERFACES (SSP MODE)

Rechargeable
Li-ion battery

Make a measurement in just a few seconds

Ready

Countdown

Measurement complete!

The device waits
for a laser beam

Automatically starts when
exposed to a laser beam

The value is displayed
until the next measurement

Adjust the wavelength and calibration

Warns you when
the device is too hot*Set the brightness
and orientation

Press PLAY

USB connector for:- Battery charging- Data transfer to PC- Internet upgrades

8-32 mounting
holes for post

Wavelength
Color LCD display withtouch screen controls

Calibration

Rugged casing
compact size
On/Offand main menu

KEY FEATURES
	POCKET-SIZE
This mid to high power laser probe is so compact itfits in your pocket!
EASY TO USE
The color LCD touchscreen allows for a friendly userinterface. You can make a measurement with justthe touch of a button!
DATA LOGGING
Save your data to the internal memory and thentransfer them to your PC over the USB connection.
FROM LOW TO HIGH POWERS
Thanks to a low noise level and high damage
threshold, the PRONTO can measure powers from
0.5 W to 250 W.
THE FLEXIBILITY TO PICK THE CALIBRATIONSYOU NEEDThe PRONTO-250-FLEX offers three calibration
options so you only pay for what you need:• Default calibration “Y”: for visible to NIR
wavelengths (248 nm to 2.5 µm)* Device may get hot, it is not recommended for handheld use when making a measurement

2 MODELS FOR ALL YOUR MEASUREMENT NEEDSPRONTO-250-FLEX
PRONTO-250-FLEX comes with 3 measurement modes and can be used ina variety of applications:Single shot power (SSP): up to 250 W
Continuous power (CWP): up to 8 W
Single shot energy (SSE): up to 25 J
PRONTO-50-W5This model has our proprietary absorber with extremely high damage
thresholds to handle tightly focused beams without damaging theabsorber.
Single shot power (SSP): up to 50 W

CONNECTIVITY• Additional calibration “C”: for CO2 lasers (10.6 µm)• Additional calibration “E”: for energy
measurements with ± 5 % accuracy

HANDS-FREE OPERATION
Place it on a flat surface or use one of the 2 threadedholes for safe use with optical stands.
SERIAL COMMANDS
Serial commands are available to let you take fullcontrol of your PRONTO from your PC.
32

gentec-eo.com/laser-power-meter

HANDS-FREE

DATA TRANSFER TO PC

PRONTOSpecifications

t

uc

ew

od
pr

N

POWER DETECTORSPRONTO-250-FLEX

PRONTO-50-W5

SSP Mode

CWP Mode

SSE Mode

MAX AVERAGE POWER/ENERGY

250 W

8W

25 J (up to 150 J for pulses >1 ms) 50 W

EFFECTIVE APERTURE

19 mm Ø19 mm ØINTERFACE

Touchscreen color LCD display

Touchscreen color LCD display

Spectral range

0.19 - 20 µm

0.19 - 10 µm

Calibrated spectral range

0.248 - 2.5 µm (default) 10.6 µm available with calibration option “C”

0.248 - 2.5 µm

Noise equivalent power/energy

10 mW

10 mW

60 mJ

4 mW

Minimum measurable power/energy 0.5 W

0.2 W

N/A

0.5 W

Exposure time5s

1.5 s response time

0.26 s

5s

Measurement accuracy± 3%± 2.5 %

± 5 % with additional
calibration "E"
Typical value as default± 3%

Min repetition period (Max pulse width)

N/A

N/A

4 s (88 ms)

N/A

Display resolution

1 mW

1 mW

10 mJ

1 mW

Measures in 5 s

Measures power continuously

Measures single-shot energy

MEASUREMENT CAPABILITY
ENERGY DETECTORS

Maximum average power density a

45 kW/cm2 (at 1064 nm, 10 W, CW)
14 kW/cm2 (at 10.6 µm, 10 W, CW)

Maximum exposure time b

6s

N/A

N/A

6s

65°C

40°C

40°C

65°C

Maximum device temperatureb

BEAM PROFILINGDAMAGE THRESHOLDS
100 kW/cm2 (at 1064 nm, 10 W, CW)

USER INTERFACE
Wavelength selection and user calibration

Wavelength selection
and user calibration

Measurement modes

Single Shot Power (SSP), Continuous Power (CWP) andSingle Shot Energy (SSE)

Single Shot Power (SSP)

Data acquisition and transfer

Yes

Yes

Display type

Touchscreen color LCD

Touchscreen color LCD

Display size

28.0 x 35.0 mm (128 x 160 pixels)

28.0 x 35.0 mm (128 x 160 pixels)

Data storage

50 000 pts

50 000 pts

Battery type

Rechargeable Li-ion

Rechargeable Li-ion

Battery life

17 hours or 4 200 measurements

17 hours or 4 200 measurements (with brightness set at 25 %)TERAHERTZ DETECTORS

Measurement controls

GENERAL SPECIFICATIONSDISPLAYS & PC INTERFACES(with brightness set at 25 %)Battery recharge via

USB port

USB port

Effective aperture

19 mm Ø19 mm ØAbsorber

H9

W5

Mounting holes (for post)

2 x 8-32

2 x 8-32

Dimensions

59W x 181.4L x 17D

59W x 181.4L x 17D

Weight

210 g

210 g

STAND-S-233

STAND-S-233

PHYSICAL CHARACTERISTICS

Compatible stand

CUSTOM / OEM PRODUCTS

ORDERING INFORMATION
Product pagea. To get all the damage thresholds, see User Manual.
b. At maximum power.
PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

33

UP25-H

25 mm Ø, 3 mW - 350 W

OUTPUT OPTIONSSMART DB15 CONNECTOR
Contains all the calibration data
ALL-IN-ONE-METER
Connects directly to a PC
Two models available: 				USB output (-INT)
RS-232 output (-IDR)
BLU WIRELESS METER
Connects via Bluetooth® to a smartphone, tablet or PC

COMPATIBLE DISPLAYS & PC INTERFACESKEY FEATURESMIRO ALTITUDE

MAESTRO

TUNER

UNO

U-LINK and P-LINK

S-LINK and M-LINK

Stand with steel post

Extension cables(4, 15, 20 or 25 m)

Fiber adaptors and connectors(FC, SC or SMA)

12V power supply

Pelican carrying case

ACCESSORIESMODULAR CONCEPT
Increase the power capability of your detector:
4 different cooling modules
HIGH PERFORMANCE
Fast rise time (1.3 s)High damage threshold (45 kW/cm2)
ENERGY MODE
Measure single shot energy up to 40 J

34gentec-eo.com/laser-power-meter

UP25-H
Specifications

UP25N-100H-H9-D0

UP25N-250F-H12-D0

UP25M-350W-H12-D0

40 W / 80 W

100 W / 200 W

250 W / 300 W

350 W f / 350 W f

EFFECTIVE APERTURE

25 mm Ø25 mm Ø25 mm Ø25 mm ØCOOLING METHOD

Convection

Heatsink

Fan-cooled

Water-cooledSpectral range

0.19 - 20 µm

0.19 - 20 µm

0.19 - 20 µm

0.19 - 20 µm

Calibrated spectral rangea

0.248 - 2.1 µm

0.248 - 2.1 µm

0.248 - 2.1 µm

0.248 - 2.1 µm

Noise equivalent power b

3 mW

3 mW

10 mW

10 mW

Rise time (nominal)(CONTINUOUS / 1 MINUTE)

POWER DETECTORSUP25N-40S-H9-D0
MAX AVERAGE POWER

MEASUREMENT CAPABILITY

1.3 s

1.3 s

1.3 s

± 2.5 %

± 2.5 %

± 2.5 %

± 2.5 %

Repeatability±0.5 %±0.5 %±0.5 %±0.5 %

Maximum measurable energy e

40 J

40 J

40 J

40 J

Noise equivalent energy b

0.2 J

0.2 J

0.2 J

0.2 J

Minimum repetition period

4.6 s

4.6 s

11.5 s11.5 sMaximum pulse width

123 ms

123 ms390 ms390 ms

± 5%

± 5%

± 5%

ENERGY DETECTORS

1.3 s

Calibration uncertainty dcEnergy mode

BEAM PROFILING

Accuracy with energy calibration option ± 5 %DAMAGE THRESHOLDS
Maximum average power density
45 kW/cm2

45 kW/cm2

45 kW/cm2

45 kW/cm2

10.6 µm, 10 W, CW

14 kW/cm

14 kW/cm

14 kW/cm

14 kW/cm2

1064 nm, 360 µs, 5 Hz

9 J/cm2

9 J/cm2

9 J/cm2

9 J/cm2

1064 nm, 7 ns, 10 Hz

1 J/cm

1 J/cm

1 J/cm2

2

2

TERAHERTZ DETECTORS1064 nm, 10 W, CW
Maximum energy density22

1 J/cm2

2

532 nm, 7 ns, 10 Hz

2

0.6 J/cm

2

0.6 J/cm

2

0.6 J/cm

0.6 J/cm2

266 nm, 7 ns, 10 Hz

0.3 J/cm2

0.3 J/cm2

0.3 J/cm2

0.3 J/cm2

Effective aperture

25 mm Ø25 mm Ø25 mm Ø25 mm ØAbsorber (high damage threshold)

H9

H9

H12

H12

Dimensions

89H x 89W x 32D mm

89H x 89W x 106D mm

92H x 92W x 117D mm

89H x 89W x 40D mm

Weight (head only)

0.68 kg

0.99 kg

1.44 kg

0.87 kg

Available output options

DB15, USB, RS-232 or Bluetooth

DB15, USB, RS-232 or Bluetooth

DB15, USB, RS-232 or Bluetooth

DB15, USB, RS-232 or BluetoothCompatible stand

STAND-S-443

STAND-S-443

STAND-S-443

STAND-S-443

PHYSICAL CHARACTERISTICSDISPLAYS & PC INTERFACESORDERING INFORMATION

Product page

CUSTOM / OEM PRODUCTS

a. Calibrations at 2.1 to 2.5 µm and 10.6 µm are available on special request.
b. Nominal value, actual value depends on electrical noise in the measurement system.
c. With anticipation.
d. Including linearity with power.
e. For 360 µs pulses. Higher pulse energy possible for long pulses (ms), less for short pulses (ns).
f. Minimum cooling flow 1.5 liters/min, water temperature ≤ 22°C, 1/8 NPT compression fittings for 1/4 inch semi-rigid tube.
Contact Gentec-EO for clean deionized water cooling module option.

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

35

UP55-H

55 mm Ø, 5 mW - 500 W

OUTPUT OPTIONSSMART DB15 CONNECTOR
Contains all the calibration data
ALL-IN-ONE-METER
Connects directly to a PC
Two models available: 				USB output (-INT)
RS-232 output (-IDR)
BLU WIRELESS METER
Connects via Bluetooth® to a smartphone, tablet or PC

COMPATIBLE DISPLAYS & PC INTERFACESKEY FEATURESMIRO ALTITUDE

MAESTRO

TUNER

UNO

U-LINK and P-LINK

S-LINK and M-LINK

Stand with steel post

Extension cables(4, 15, 20 or 25 m)

Fiber adaptors and connectors(FC, SC or SMA)

3-Port fiber cylinder withadaptors and plug

12V power supply

Pelican carrying case

ACCESSORIESMODULAR CONCEPT
Increase the power capability of your detector:
4 different cooling modules
HIGH PERFORMANCE
Fast rise time (2 s)High damage threshold (45 kW/cm2)
COMPACT DESIGN
Only 32 mm thick (40S model)
ENERGY MODE
Measure single shot energy up to 200 J

36

gentec-eo.com/laser-power-meterUP55-H
Specifications

UP55N-100H-H9-D0

UP55N-300F-H12-D0

UP55M-500W-H12-D0

MAX AVERAGE POWER

40 W / 80 W

100 W / 200 W

300 W / 300 W

500 W f / 500 W f

EFFECTIVE APERTURE

55 mm Ø55 mm Ø55 mm Ø55 mm ØCOOLING METHOD

Convection

Heatsink

Fan-cooled

Water-cooledSpectral range

0.19 - 20 µm

0.19 - 20 µm

0.19 - 20 µm

0.19 - 20 µm

Calibrated spectral range a

0.248 - 2.1 µm

0.248 - 2.1 µm

0.248 - 2.1 µm

0.248 - 2.1 µm

Noise equivalent power

5 mW

5 mW

15 mW

15 mW

(CONTINUOUS / 1 MINUTE)

POWER DETECTORSUP55N-40S-H9-D0

MEASUREMENT CAPABILITY

2s

2s

2s

2s

Calibration uncertainty d± 2.5 %

± 2.5 %

± 2.5 %

± 2.5 %

Repeatability±0.5 %±0.5 %±0.5 %±0.5 %

Maximum measurable energy e

200 J

200 J

200 J

200 J

Noise equivalent energy

0.25 J

0.25 J

0.25 J

0.25 J

Minimum repetition period

11.1 s11.1 s

12 s

12 s

Maximum pulse width

433 ms

433 ms

430 ms

430 ms

± 5%

± 5%

± 5%

c

ENERGY DETECTORS

Rise time (nominal)b

Energy modeb

BEAM PROFILING	Accuracy with energy calibration option ± 5 %DAMAGE THRESHOLDS
Maximum average power density
45 kW/cm2

45 kW/cm2

45 kW/cm2

45 kW/cm2

10.6 µm, 10 W, CW

14 kW/cm

14 kW/cm

14 kW/cm

14 kW/cm2

1064 nm, 360 µs, 5 Hz

9 J/cm2

9 J/cm2

9 J/cm2

9 J/cm2

1064 nm, 7 ns, 10 Hz

1 J/cm

1 J/cm

1 J/cm

1 J/cm2

532 nm, 7 ns, 10 Hz

0.6 J/cm2

0.6 J/cm2

0.6 J/cm2

0.6 J/cm2

266 nm, 7 ns, 10 Hz

0.3 J/cm2

0.3 J/cm2

0.3 J/cm2

0.3 J/cm2

Effective aperture

55 mm Ø55 mm Ø55 mm Ø55 mm ØAbsorber (high damage threshold)

H9

H9

H12

H12

Dimensions

89H x 89W x 32D mm

89H x 89W x 106D mm

92H x 92W x 117D mm

89H x 89W x 40D mm

Weight (head only)

0.62 kg

0.93 kg

1.41 kg

0.81 kg

DB15, USB, RS-232
or Bluetooth

DB15, USB, RS-232
or Bluetooth

DB15, USB, RS-232
or Bluetooth

DB15, USB, RS-232
or Bluetooth

STAND-S-443

STAND-S-443

STAND-S-443

STAND-S-443

2

2

2

TERAHERTZ DETECTORS1064 nm, 10 W, CW
Maximum energy density

2

2

2

PHYSICAL CHARACTERISTICS

Compatible stand

DISPLAYS & PC INTERFACESORDERING INFORMATION
Available output options

Product page

CUSTOM / OEM PRODUCTS

a. Calibrations at 2.1 to 2.5 µm and 10.6 µm are available on special request.
b. Nominal value, actual value depends on electrical noise in the measurement system.
c. With anticipation.
d. Including linearity with power.
e. For 360 µs pulses. Higher pulse energy possible for long pulses (ms), less for short pulses (ns).
f. Minimum cooling flow 1.5 liters/min, water temperature ≤ 22°C, 1/8 NPT compression fittings for 1/4 inch semi-rigid tube. Contact Gentec-EO for clean deionized water cooling module option.
PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

37

UP55-HD

55 mm Ø, 45 mW - 2500 W

OUTPUT OPTIONSSMART DB15 CONNECTOR
Contains all the calibration data
ALL-IN-ONE-METER
Connects directly to a PC
Two models available: 				USB output (-INT)
RS-232 output (-IDR)
BLU WIRELESS METER
Connects via Bluetooth® to a smartphone, tablet or PC

COMPATIBLE DISPLAYS & PC INTERFACES
600 W

2.5 kW

MIRO ALTITUDE

MAESTRO

TUNER

UNO

U-LINK and P-LINK

S-LINK and M-LINK

Stand with steel post

Extension cables(4, 15, 20 or 25 m)

Fiber adaptors and connectors(FC, SC or SMA)

3-Port fiber cylinder withadaptors and plug

12V power supply

Pelican carrying case

700 W

KEY FEATURESACCESSORIESHIGH DENSITY ABSORBER
The HD absorber is the strongest on the market
for use at high powers, presenting both highaverage power handling and high power density
capabilities
UP55G-600F-HD - NO NEED FOR
WATER COOLING
Unique on the market, measure 600 W ofcontinuous power WITHOUT THE NEED FOR
WATER COOLING. Just plug the fan and you areready to go!
	UP55M-700W-HD - FAST AND COMPACT
A very compact detector that measures up
to 700 W of continuous power.
	UP55C-2.5KW-HD - PERFORMANCE AND
SPEED AT A LOW PRICE
Measures both very low and very high powers (upto 2500W) with a fast response time. A compact
and versatile detector that is more affordable thanany other high power solution on the market.
38

gentec-eo.com/laser-power-meterUP55-HDSpecifications

UP55C-2.5KW-HD-D0

600 W / 600 W

700 W f / 700 W f

2500 W f / 2500 W f

EFFECTIVE APERTURE

55 mm Ø55 mm Ø55 mm ØCOOLING METHOD

Fan-cooled

Water-cooled

Water-cooled

MEASUREMENT CAPABILITY
Spectral range

0.19 - 20 µm

0.19 - 20 µm

0.19 - 20 µm

Calibrated spectral range a

0.248 - 2.1 µm

0.248 - 2.1 µm

0.248 - 2.1 µm

Noise equivalent power b

45 mW

45 mW

200 mW

Rise time (nominal) c

2,8 s

2.8 s

3.5 s

Calibration uncertainty d± 2.5 %

± 2.5 %

± 2.5 %

Repeatability±0.5 %±0.5 %±0.5 %

Maximum measurable energy d

200 J

200 J

---

Noise equivalent energy(CONTINUOUS / 1 MINUTE)ENERGY DETECTORS

UP55M-700W-HD-D0

POWER DETECTORSUP55G-600F-HD-D0
MAX AVERAGE POWER

Energy mode
0.25 J

---

12 s

12 s

---

Maximum pulse width

430 ms

430 ms

---Accuracy with energy calibration option

± 5%

± 5%

---

1064 nm, 10 W, CW

45 kW/cm2

45 kW/cm2

45 kW/cm2

1064 nm, 500 W, CW

8 kW/cm2

8 kW/cm2

9 kW/cm2

1064 nm, 2500 W, CW

---

---

6 kW/cm2

10.6 μm, 500 W, CW

---

---

4.5 kW/cm2

10.6 μm, 1500 W, CW

---

---

3.5 kW/cm2

10.6 μm, 2500 W, CW

---

---

3.0 kW/cm2

BEAM PROFILING

0.25 J

Minimum repetition periodb

DAMAGE THRESHOLDS
Maximum average power density

TERAHERTZ DETECTORS

Maximum energy density
9 J/cm2

9 J/cm2

9 J/cm2

1064 nm, 7 ns, 10 Hz

1 J/cm

1 J/cm

1 J/cm2

532 nm, 7 ns, 10 Hz

0.6 J/cm2

0.6 J/cm2

0.6 J/cm2

266 nm, 7 ns, 10 Hz

0.3 J/cm2

0.3 J/cm2

0.3 J/cm2

PHYSICAL CHARACTERISTICS
Effective aperture2

2

55 mm Ø55 mm Ø55 mm ØAbsorber (high damage threshold)

HD

HD

HD

Dimensions

120H x 120W x 135D mm

89H x 89W x 40D mm

116H x 116W x 37D mm

Weight (head only)

2.75 kg

0.90 kg

3.3 kgDB15, USB, RS-232 or Bluetooth

DB15, USB, RS-232 or Bluetooth

DB15, USB, RS-232 or Bluetooth

STAND-S-443-C

STAND-S-443-C

STAND-S-443-C

ORDERING INFORMATION
Available output options
Compatible stand

DISPLAYS & PC INTERFACES

1064 nm, 360 µs, 5 Hz

Product page

CUSTOM / OEM PRODUCTS

a. Calibrations at 2.1 to 2.5 µm and 10.6 µm are available on special request.
b. Nominal value, actual value depends on electrical noise in the measurement system.
c. With anticipation.
d. Including linearity with power.
e. For 360 μs pulses. Higher pulse energy possible for long pulses (ms), less for short pulses (ns).
f. Minimum cooling flow 1.5 l/m (UP55M-700W-HD) or 3 l/m (UP55C-2.5KW-HD), water temperature ≤22°C, 1/8 NPT compression fittings for 1/4 inch semi-rigid tube.
Contact Gentec-EO for clean deionized water cooling module option.
PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

39

UP19-W

19 mm Ø, 1 mW - 85 W, 100 kW/cm2

OUTPUT OPTIONSSMART DB15 CONNECTOR
Contains all the calibration data
ALL-IN-ONE-METER
Connects directly to a PC
Two models available: 				USB output (-INT)
RS-232 output (-IDR)
BLU WIRELESS METER
Connects via Bluetooth® to a smartphone, tablet or PC

COMPATIBLE DISPLAYS & PC INTERFACESKEY FEATURESMIRO ALTITUDE

MAESTRO

TUNER

UNO

U-LINK and P-LINK

S-LINK and M-LINK

Stand with steel post

Extension cables(4, 15, 20 or 25 m)

12V power supply

Pelican carrying case

Isolation tube

Fiber adaptors & connectors(FC, ST and SMA)ACCESSORIESMODULAR CONCEPT
Increase the power capability of your detector:
4 different cooling modules
VERY HIGH DAMAGE THRESHOLD
100 kW/cm2 in average power density
COMPACT DESIGN
Only 21 mm thick (15S model)
ENERGY MODE
Measure single shot energy up to 200 J

40gentec-eo.com/laser-power-meter

UP19-W
Specifications

UP19K-30H-W5-D0

UP19K-50L-W5-D0

UP19K-50F-W5-D0

15 W / 30 W

30 W / 60 W

50 W / 85 W

50 W / 85 W

EFFECTIVE APERTURE

19 mm Ø19 mm Ø19 mm Ø19 mm ØCOOLING METHOD

Convection

Heatsink

Large heatsink

Fan-cooledSpectral range

0.19 - 10.0 µm

0.19 - 10.0 µm

0.19 - 10.0 µm

0.19 - 10.0 µm

Calibrated spectral range a

0.248 - 2.1 µm

0.248 - 2.1 µm

0.248 - 2.1 µm

0.248 - 2.1 µm

Noise equivalent power b

1 mW

1 mW

1 mW

1 mW

Rise time (nominal)(CONTINUOUS / 1 MINUTE)

POWER DETECTORSUP19K-15S-W5-D0
MAX AVERAGE POWER

MEASUREMENT CAPABILITY

1.4 s

1.4 s

1.4 s

± 2.5 %

± 2.5 %

± 2.5 %

± 2.5 %

Repeatability±0.5 %±0.5 %±0.5 %±0.5 %

Maximum measurable energy e

200 J

200 J

200 J

200 J

Noise equivalent energy b

0.02 J

0.02 J

0.02 J

0.02 J

Minimum repetition period

5s

5s

5s

5s

Maximum pulse width

133 ms

133 ms

133 ms

133 ms

± 5%

± 5%

± 5%

100 kW/cm2

100 kW/cm2

100 kW/cm2

100 kW/cm2

1064 nm, 150 µs, 10 Hz

100 J/cm2

100 J/cm2

100 J/cm2

100 J/cm2

1064 nm, 7 ns, 10 Hz

1.1 J/cm

1.1 J/cm

1.1 J/cm

1.1 J/cm2

532 nm, 7 ns, 10 Hz

1.1 J/cm2

248 nm, 26 ns, 10 Hz

0.7 J/cm

0.7 J/cm

0.7 J/cm

0.7 J/cm2

Effective aperture

19 mm Ø19 mm Ø19 mm Ø19 mm ØAbsorber (high damage threshold)

W5

W5

W5

W5

Dimensions

50H x 50W
x 20.6D mm

50H x 50W
x 56.3D mm

76.2H x 76.2W
x 73.6D mm

50H x 50W
x 63D mm

Weight (head only)

0.16 kg

0.21 kg

0.48 kg

0.25 kg

DB15, USB, RS-232
or Bluetooth

DB15, USB, RS-232
or Bluetooth

DB15, USB or RS-232

DB15, USB, RS-232
or Bluetooth

STAND-S-233

STAND-S-233

STAND-S-233

STAND-S-233

ENERGY DETECTORS

1.4 s

Calibration uncertainty dcEnergy mode

BEAM PROFILING	Accuracy with energy calibration option ± 5 %DAMAGE THRESHOLDS
Maximum average power density f
Maximum energy density22

1.1 J/cm2
2

1.1 J/cm2
2

TERAHERTZ DETECTORS

2

1.1 J/cm2
2

PHYSICAL CHARACTERISTICS

Compatible stand

DISPLAYS & PC INTERFACESORDERING INFORMATION
Available output options

Product page

CUSTOM / OEM PRODUCTS

a. Calibration at 2.1 to 2.5 µm is available on special request.
b. Nominal value, actual value depends on electrical noise in the measurement system.
c. With anticipation.
d. Including linearity with power.
e. For 150 µs pulses. Higher pulse energy possible for long pulses (ms), less for short pulses (ns).
f. At 1064 nm, 10 W CW.

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

41

UP50-W

50 mm Ø, 5 mW - 85 W, 100 kW/cm2

OUTPUT OPTIONSSMART DB15 CONNECTOR
Contains all the calibration data
ALL-IN-ONE-METER
Connects directly to a PC
Two models available: 				USB output (-INT)
RS-232 output (-IDR)
BLU WIRELESS METER
Connects via Bluetooth® to a smartphone, tablet or PC

COMPATIBLE DISPLAYS & PC INTERFACESKEY FEATURESMIRO ALTITUDE

MAESTRO

TUNER

UNO

U-LINK and P-LINK

S-LINK and M-LINK

Stand with steel post

Extension cables(4, 15, 20 or 25 m)

Fiber adaptors and connectors(FC, SC or SMA)

3-Port fiber cylinder withadaptors and plug

12V power supply

Pelican carrying case

ACCESSORIESMODULAR CONCEPT
Increase the power capability of your detector:
3 different cooling modules
VERY HIGH DAMAGE THRESHOLD
100 kW/cm2 in average power density
VERY LARGE APERTURE
50 mm Ø effective aperture, perfect for large beams
HIGHEST ENERGY READINGS IN THE SERIES
Measure single shot energy up to 500 J

42

gentec-eo.com/laser-power-meter

UP50-W
Specifications

UP50N-50H-W9-D0

UP50N-50F-W9-D0

40 W / 80 W

50 W / 85 W

50 W / 85 W

EFFECTIVE APERTURE

50 mm Ø50 mm Ø50 mm ØCOOLING METHOD

Convection

Heatsink

Fan-cooledSpectral range

0.19 - 10.0 µm

0.19 - 10.0 µm

0.19 - 10.0 µm

Calibrated spectral range a

0.248 - 2.1 µm

0.248 - 2.1 µm

0.248 - 2.1 µm

Noise equivalent power b

5 mW

5 mW

5 mW

Rise time (nominal)(CONTINUOUS / 1 MINUTE)

POWER DETECTORSUP50N-40S-W9-D0
MAX AVERAGE POWER

MEASUREMENT CAPABILITY

3.5 s

3.5 s

± 2.5 %

± 2.5 %

± 2.5 %

Repeatability±0.5 %±0.5 %±0.5 %

Maximum measurable energy e

500 J

500 J

500 J

Noise equivalent energy b

0.25 J

0.25 J

0.25 J

Minimum repetition period

11.1 s11.1 s11.1 sMaximum pulse width

467 ms

467 ms

467 ms± 5%± 5%

100 kW/cm2

100 kW/cm2

100 kW/cm2

1064 nm, 150 µs, 5 Hz

100 J/cm2

100 J/cm2

100 J/cm2

1064 nm, 7 ns, 10 Hz

1.1 J/cm

1.1 J/cm

1.1 J/cm2

532 nm, 7 ns, 10 Hz

1.1 J/cm2

248 nm, 26 ns, 10 Hz

ENERGY DETECTORS

3.5 s

Calibration uncertainty dcEnergy mode

BEAM PROFILING	Accuracy with energy calibration option ± 5 %DAMAGE THRESHOLDS
Maximum average power density f
Maximum energy density2

1.1 J/cm2

TERAHERTZ DETECTORS

2

1.1 J/cm2

2

0.7 J/cm

2

0.7 J/cm

0.7 J/cm2

Effective aperture

50 mm Ø50 mm Ø50 mm ØAbsorber (high damage threshold)

W9

W9

W9

Dimensions

89H x 89W x 38D mm

89H x 89W x 109D mm

92H x 92W x 116D mm

Weight (head only)

0.62 kg

0.93 kg

1.38 kg

Available output options

DB15, USB, RS-232 or Bluetooth

DB15, USB, RS-232 or Bluetooth

DB15, USB, RS-232 or BluetoothCompatible stand

STAND-S-443

STAND-S-443

STAND-S-443

PHYSICAL CHARACTERISTICSDISPLAYS & PC INTERFACESORDERING INFORMATION

Product page

CUSTOM / OEM PRODUCTS

a. Calibration at 2.1 to 2.5 µm is available on special request.
b. Nominal value, actual value depends on electrical noise in the measurement system.
c. With anticipation.
d. Including linearity with power.
e. For 360 µs pulses. Higher pulse energy possible for long pulses (ms), less for short pulses (ns).
f. At 1064 nm, 10 W CW.

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

43

UP16-QED

16 mm Ø, 4 mW - 100 W, volume absorber

OUTPUT OPTIONSSMART DB15 CONNECTOR
Contains all the calibration data
ALL-IN-ONE-METER
Connects directly to a PC
Two models available: 				USB output (-INT)
RS-232 output (-IDR)
BLU WIRELESS METER
Connects via Bluetooth® to a smartphone, tablet or PC

COMPATIBLE DISPLAYS & PC INTERFACESKEY FEATURESMIRO ALTITUDE

MAESTRO

TUNER

UNO

U-LINK and P-LINK

S-LINK and M-LINK

Extension cables(4, 15, 20 or 25 m)

Pelican carrying case

ACCESSORIESMODULAR CONCEPT
Increase the power capability of your detector:
3 different cooling modules
HIGH PEAK POWER VOLUME ABSORBER
Perfect for pulsed beams with high energy density
COMPACT DESIGN
Only 24 mm thick (15S model)
ENERGY MODE
Measure single shot energy up to 500 J

44

gentec-eo.com/laser-power-meterStand with steel post

UP16-QED
Specifications

UP16K-30H-QED-D0

UP16K-100W-QED-D0

MAX AVERAGE POWER

15 W / 20 W

30 W / 35 W

100 W / 100 W

EFFECTIVE APERTURE

16 mm Ø16 mm Ø16 mm ØCOOLING METHOD

Convection

Heatsink

Water-cooledSpectral range

0.266 - 2.5 µm

0.266 - 2.5 µm

0.266 - 2.5 µm

Calibrated spectral range a

0.532 - 2.1 µm

0.532 - 2.1 µm

0.532 - 2.1 µm

Noise equivalent power(CONTINUOUS / 1 MINUTE)

POWER DETECTORSUP16K-15S-QED-D0

MEASUREMENT CAPABILITY

4 mW

4 mW

2.5 s

2.5 s

2.5 s

Calibration uncertainty d± 2.5 %

± 2.5 %

± 2.5 %

Repeatability±0.5 %±0.5 %±0.5 %

Maximum measurable energy e

500 J

500J

500 J

Noise equivalent energy

ENERGY DETECTORS

4 mW

Rise time (nominal) cbEnergy mode
60 mJ

60 mJ

4s

4s

4s

Maximum pulse width

61 ms

61 ms

61 ms

Accuracy with energy calibration option ± 5 %

± 5%

± 5%

100 kW/cm2

100 kW/cm2

100 kW/cm2

1064 nm, 360 µs, 5 Hz

300 J/cm2

300 J/cm2

300 J/cm2

1064 nm, 7 ns, 10 Hz

8 J/cm2

8 J/cm2

8 J/cm2

532 nm, 7 ns, 10 Hz2

6 J/cm2

6 J/cm

6 J/cm2

266 nm, 7 ns, 10 Hz

2

1 J/cm2

1 J/cm

1 J/cm2

Effective aperture

16 mm Ø16 mm Ø16 mm ØAbsorber (volume absorber)

QED

QED

QED

Dimensions

50H x 50W x 23.6D mm

50H x 50W x 59D mm

50H x 50W x 38D

Weight (head only)

0.16 kg

0.21 kg

0.24 kg

Available output options

DB15, USB, RS-232 or Bluetooth

DB15, USB, RS-232 or Bluetooth

DB15, USB, RS-232 or BluetoothCompatible stand

STAND-S-233

STAND-S-233

STAND-S-233

BEAM PROFILING

60 mJ

Minimum repetition periodb

DAMAGE THRESHOLDS
Maximum average power density f
Maximum energy density

TERAHERTZ DETECTORSPHYSICAL CHARACTERISTICSDISPLAYS & PC INTERFACESORDERING INFORMATION

Product page

CUSTOM / OEM PRODUCTS

a. Calibration at 2.1 to 2.5 µm is available on special request.
b. Nominal value, actual value depends on electrical noise in the measurement system.
c. With anticipation.
d. Including linearity with power.
e. For 360 µs pulses. Higher pulse energy possible for long pulses (ms), less for short pulses (ns).
f. At 1064 nm, 10 W CW.

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

45

UP52-QED

52 mm Ø, 15 mW - 300 W, volume absorber

OUTPUT OPTIONSSMART DB15 CONNECTOR
Contains all the calibration data
ALL-IN-ONE-METER
Connects directly to a PC
Two models available: 				USB output (-INT)
RS-232 output (-IDR)
BLU WIRELESS METER
Connects via Bluetooth® to a smartphone, tablet or PC

COMPATIBLE DISPLAYS & PC INTERFACESKEY FEATURESMIRO ALTITUDE

MAESTRO

TUNER

UNO

U-LINK and P-LINK

S-LINK and M-LINK

Extension cables(4, 15, 20 or 25 m)

12V power supply

ACCESSORIESMODULAR CONCEPT
Increase the power capability of your detector:
4 different cooling modules
HIGH PEAK POWER VOLUME ABSORBER
Perfect for pulsed beams with high energy density
LARGE APERTURE
52 mm Ø aperture accomodates large beams

Stand with steel post

HIGH AVERAGE POWER
Up to 300 W of continuous power with thewater-cooled unit
ENERGY MODE
Measure single shot energy up to 1000 J

46

gentec-eo.com/laser-power-meter

Pelican carrying case

UP52-QED
Specifications

UP52N-100H-QED-D0

UP52N-150F-QED-D0

UP52M-300W-QED-D0

50 W / 50 W

100 W / 100 W

150 W / 150 W

300 W f / 300 W f

EFFECTIVE APERTURE

52 mm Ø52 mm Ø52 mm Ø52 mm ØCOOLING METHOD

Convection

Heatsink

Fan-cooled

Water-cooledSpectral range

0.266 - 2.5 µm

0.266 - 2.5 µm

0.266 - 2.5 µm

0.266 - 2.5 µm

Calibrated spectral range a

0.300 - 2.1 µm

0.300 - 2.1 µm

0.300 - 2.1 µm

0.300 - 2.1 µm

Noise equivalent power b

15 mW

15 mW

15 mW

15 mW

Rise time (nominal)(CONTINUOUS / 1 MINUTE)

POWER DETECTORSUP52N-50S-QED-D0
MAX AVERAGE POWER

MEASUREMENT CAPABILITY

4s

4s

4s

± 2.5 %

± 2.5 %

± 2.5 %

± 2.5 %

Repeatability±0.5 %±0.5 %±0.5 %±0.5 %

Maximum measurable energy e1000 J

1000 J

1000 J

1000 J

Noise equivalent energy b

250 mJ

250 mJ

250 mJ

250 mJ

Minimum repetition period

9s

9s

9s

9s

Maximum pulse width

371 ms

371 ms

371 ms

371 ms

± 5%

± 5%

± 5%

100 kW/cm2

100 kW/cm2

100 kW/cm2

100 kW/cm2

1064 nm, 360 µs, 5 Hz

300 J/cm2

300 J/cm2

300 J/cm2

300 J/cm2

1064 nm, 7 ns, 10 Hz2

8 J/cm2

8 J/cm2

8 J/cm

6 J/cm2

532 nm, 7 ns, 10 Hz

6 J/cm2

6 J/cm2

6 J/cm2

6 J/cm2

266 nm, 7 ns, 10 Hz

1 J/cm

1 J/cm

1 J/cm

1 J/cm2

Effective aperture

52 mm Ø52 mm Ø52 mm Ø52 mm ØAbsorber (volume absorber)

QED

QED

QED

QED

Dimensions

89H x 89W x 32D mm

89H x 89W x 106D mm

89H x 89W x 116D mm

89H x 89W x 43D mm

Weight (head only)

0.62 kg

0.93 kg

1.41 kg

0.84 kg

DB15, USB, RS-232
or Bluetooth

DB15, USB, RS-232
or Bluetooth

DB15, USB, RS-232
or Bluetooth

DB15, USB, RS-232
or Bluetooth

STAND-S-443

STAND-S-443

STAND-S-443

STAND-S-443

ENERGY DETECTORS

4s

Calibration uncertainty dcEnergy mode

BEAM PROFILING

Accuracy with energy calibration option ± 5 %DAMAGE THRESHOLDS
Maximum average power density g
Maximum energy density

2

2

TERAHERTZ DETECTORS

2

PHYSICAL CHARACTERISTICS

Compatible stand

DISPLAYS & PC INTERFACESORDERING INFORMATION
Available output options

Product page

CUSTOM / OEM PRODUCTS

a. Calibrations at 2.1 to 2.5 µm and 10.6 µm are available on special request.
b. Nominal value, actual value depends on electrical noise in the measurement system.
c. With anticipation.
d. Including linearity with power.
e. For 360 µs pulses. Higher pulse energy possible for long pulses (ms), less for short pulses (ns).
f. Minimum cooling flow 1 liters/min, water temperature ≤ 22°C, 1/8 NPT compression fittings for 1/4 inch semi-rigid tube.
Contact Gentec-EO for clean deionized water cooling module option.
g. At 1064 nm, 10 W CW.

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

47

IS

Integrating sphere detector for laser power measurement up to 1 kW

OUTPUT OPTIONS
ALL-IN-ONE-METER
Connects directly to a PC
Two models available: 		USB output (-INT)
RS-232 output (-IDR)

COMPATIBLE PC INTERFACESINTEGRAKEY FEATURESACCESSORIESFASTEST RESPONSE
With its silicon sensor, the integrating sphere is
as fast as a photodiode.
WIDE POWER RANGE
Very low noise level = wide power range
with just one device

Stand with delrin post

HIGH AVERAGE POWER
Measure up to 1000 W of continuous power.
RESISTANT COATING
Our proprietary coating is designed to be strong.
Its damage thresholds are orders of magnitude
higher than any other “white” coatings on themarket.
PRECISE CALIBRATIONThe IS detectors have a NIST-traceablecalibration for the entire calibrated spectral
range.

48

gentec-eo.com/laser-power-meter

Isolation tube
(for IS12L only)

Fiber adaptors & connectors(for IS12L only)

Pelican carrying case

IS

Specifications

IS50A-1KW-RSI-INT-D0

MAXIMUM AVERAGE POWER

9W

1000 W

EFFECTIVE APERTURE

12 mm Ø50 mm ØCOOLING METHOD

Convection

Water

Spectral range

340 - 1100 nm

340 - 1100 nm

Calibrated spectral range

400 - 1070 nm

400 - 1070 nm

Maximum average power

9W

1000 W

Noise equivalent power a

1 µW at 1070 nm

10 µW at 1070 nm

Maximum divergence

10° (half-angle)

10° (half-angle)

Maximum incidence angle± 10°± 25° for beam diameter < Ø 12mm
± 5° for beam diameter > Ø 12mm

Typical rise time

< 0.2 s< 0.2 sSampling rate

15 Hz

15 Hz

Calibration uncertainty± 5.0 % (400 - 499 nm)± 3.5 % (500 - 1069 nm)± 2.5 % (1070 nm)± 5.0 % (400 - 499 nm)± 3.5 % (500 - 1069 nm)± 2.5 % (1070 nm)

Back reflections b

6%

12 %

Linearity with power

± 1%

± 1%

Maximum average power density c

2 kW/cm2

5 kW/cm2

Maximum energy density d

400 mJ/cm2

400 mJ/cm2

Effective aperture

12 mm Ø50 mm ØMounting thread

SM1

SM2

Sphere inner diameter

50 mm Ø100 mm Ø

SensorSiliconSilicon

Dimensions

66H x 78W x 66D mm

127H x 140W x 115D mm

Weight0.75 kg

4 kg

Available output options

USB or RS-232

USB or RS-232

Compatible stand

STAND-S-443

STAND-S-443-C

POWER DETECTORS

IS12L-9S-RSI-INT-D0

MEASUREMENT CAPABILITY
ENERGY DETECTORSBEAM PROFILINGDAMAGE THRESHOLDSTERAHERTZ DETECTORSPHYSICAL CHARACTERISTICSDISPLAYS & PC INTERFACESORDERING INFORMATION

Product page

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

CUSTOM / OEM PRODUCTS

a. Nominal value. Actual value depends on environmental electromagnetic interference and wavelength.
b. The backscattered power (also known as back reflections) is concentrated in a cone with an apex located at the back of the sphere. For IS12, the cone has a 7.5-degree half-angle.
For IS50, the cone has a 15-degree half-angle.
c. At 1064 - 1070 nm, CW.
d. At 1064 - 1070 nm, 7 ns.

.com

49

HP60

60 mm Ø with cone reflector, 300 W - 15 000 W

OUTPUT OPTIONSSMART DB15 CONNECTOR
Contains all the calibration data
USB PORT
Connects directly to a PC
Included in all HP models
BLU WIRELESS METER
Connects via Bluetooth to a PC

COMPATIBLE DISPLAYS & PC INTERFACESMIRO ALTITUDE

MAESTRO

TUNER

Stand with steel post

Extension cables(4, 15, 20 or 25 m)*

5 m USB cable(Included)

Water filter(Metric: 202984, Imperial: 202990)

Pelican carrying caseUNO

KEY FEATURESHIGH POWER HANDLING
Handles up to 15 kW of continuous power. Custom
models available for higher powers. The new HP60A15KW-GD-QBH is designed for use with QB/QBH highpower fibers.
LOW BACK REFLECTIONS
The cone reflector traps most of the incident laser power
inside the detector head. With its TUBE extension, theHP60A-15KW-GD-TUBE has the lowest back reflection
rating: under 2 %.

ACCESSORIES	AVAILABLE WITH YAG AND CO2 CALIBRATIONSAll HP models can be calibrated at YAG and CO2
wavelengths with a calibration uncertainty of ± 5 %DIRECT USB CONNECTION TO A PC
Each head comes with both a DB15 connector(for use with a Gentec-EO display device) and a USB
output for direct connection to a PC
TRACK WATER PARAMETERS
Water flow and temperature are monitored in real timeand displayed continuously
50

gentec-eo.com/laser-power-meter*A USB power adaptor will be necessary if the HP is used with a DB15 extension cable.

HP60
Specifications

HP60A-15KW-GD

HP60A-15KW-GD-TUBE HP60A-15KW-GD-QBH

MAX AVERAGE POWER

10 kW

15 kW

15 kW

15 kW

EFFECTIVE APERTURE

60 mm Ø60 mm Ø70 mm ØQB/QBH fiber adaptor

COOLING METHOD

Water-cooled

Water-cooled

Water-cooled

Water-cooledSpectral range

0.8 - 12 μm

0.8 - 12 μm

0.8 - 12 μm

0.8 - 12 μm

Calibrated spectral range a

0.8 - 2.1 μm

0.8 - 2.1 μm

0.8 - 2.1 μm

0.8 - 2.1 μm

Noise equivalent power

10 W

15 W

15 W

15 W

POWER DETECTORS

HP60A-10KW-GD

MEASUREMENT CAPABILITY

Minimum average power

500 W

500 W

500 W

12 s

15 s

15 s

15 s

Back reflections

10 %5 - 10 %1 - 2%

1 - 2%

Calibration uncertainty± 5 % at 1064 nm & 1070 nm± 5 % at 1064 nm & 1070 nm± 5 % at 1064 nm & 1070 nm± 5 % at 1064 nm & 1070 nm

Repeatability

± 2%

± 2%

± 2%

± 2%

Linearity with power

± 2%

± 2%

± 2%

± 2%

Linearity with beam diameter± 2.0 %

± 2.5 %

± 2.5 %

± 2.5 %

Linearity with beam position d± 3.0 %

± 4.0 %

± 4.0 %

± 4.0 %

1 kW

70 kW/cm2

70 kW/cm2

70 kW/cm2

70 kW/cm2

5 kW

35 kW/cm2

35 kW/cm2

35 kW/cm2

35 kW/cm2

10 kW

20 kW/cm

2

20 kW/cm

2

20 kW/cm

20 kW/cm2

10 kW/cm2

10 kW/cm2

10 kW/cm2

BEAM PROFILING

300 W

Rise time (nominal)c

ENERGY DETECTORSb

DAMAGE THRESHOLDS
Maximum average power density e

2

TERAHERTZ DETECTORS

15 kW
PHYSICAL CHARACTERISTICS
Effective aperture

60 mm Ø60 mm Ø70 mm Ø tube aperture

QB/QBH fiber adaptor

Absorber

GD (cone reflector)

GD (cone reflector)

GD (cone reflector)

GD (cone reflector)

Required cooling flow f(6 - 8) LPM < ± 1 LPM/min(8 - 10) LPM < ± 1 LPM/min(8 - 10) LPM < ± 1 LPM/min(8 - 10) LPM < ± 1 LPM/min

Temperature range

15 - 25 °C

15 - 25 °C

15 - 25 °C

15 - 25 °C

Rate of temperature change< ± 3°C/min< ± 3°C/min< ± 3°C/min< ± 3°C/min

Maximum water pressure (input)

413 kPa (60 psi)

413 kPa (60 psi)

413 kPa (60 psi)

413 kPa (60 psi)

Dimensions

127H x 127W x 95D mm

153H x 153W x 97D mm

153H x 153W x 302D mm

153H x 153W x 302D mm

Weight

6 kg

10 kg

15 kg

15 kg

Available output options

DB15 & USB or Bluetooth & USB

DB15 & USB or Bluetooth & USB

DB15 & USB or Bluetooth & USB

DB15 & USB or Bluetooth & USB

Compatible stand

STAND-S-443-C

2x STAND-S-443-C

3x STAND-S-443-C

3x STAND-S-443-C

Cooling water

DISPLAYS & PC INTERFACESORDERING INFORMATION

Product page

CUSTOM / OEM PRODUCTS

a. Calibrations at 2.1 to 2.5 µm and 10.6 µm are available on special request.
b. Nominal value, actual value depends on electrical noise in the measurement system.
c. For lower powers, call your Gentec-EO representative.
d. For a beam size of 20 % of the aperture area, moved across 80 % of the aperture area.
e. At 1064 nm, 1.07-1.08 μm and 10.6 μm, for beams < 50 mm Ø.
f. > 1 min. contact Gentec-EO for deionized water cooling module option.

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

51

HP100Up to 125 x 125 mm, 100 W - 15 kW

OUTPUT OPTIONSSMART DB15 CONNECTOR
Contains all the calibration data
USB PORT
Connects directly to a PC
Included in all HP models
BLU WIRELESS METER
Connects via Bluetooth to a PC

COMPATIBLE DISPLAYS & PC INTERFACESMIRO ALTITUDE

MAESTRO

TUNER

Stand with steel post

Extension cables(4, 15, 20 or 25 m)*

5 m USB cable(Included)

Water filter(Metric: 202984, Imperial: 202990)

Pelican carrying caseUNO

KEY FEATURESACCESSORIESHIGH POWER HANDLING
Handles up to 15 kW of continuous power with ourstandard models. Custom models available for higher
powers (See SUPER HP)
LARGE APERTURE
Our standard HP models have very large effectives
apertures to accomodate large laser beams. Larger
apertures with various shapes are available uponrequest (See SUPER HP)
AVAILABLE WITH YAG AND CO2 CALIBRATIONSAll HP Models can be calibrated at YAG and CO2
wavelengths with a calibration uncertainty of ± 5 %DIRECT USB CONNECTION TO A PC
Each head comes with both a DB15 connector(for use with a Gentec-EO display device) and a USB
output for direct connection to a PC
TRACK WATER PARAMETERS
Water flow and temperature are monitored in real timeand displayed continuously*A USB power adaptor will be necessary if the HP is used with a DB15 extension cable.

52

gentec-eo.com/laser-power-meterHP100Specifications

HP100A-12KW-HD HP100A-12KWHD-TUBE

HP125A-15KW-HD HP125A-15KWHD-TUBE

12 000 W

12 000 W

15 000 W

15 000 W

70 mm Ø100 mm Ø

70 mm Ø125 x 125 mm

70 mm ØCOOLING METHOD

Water-cooled

Water-cooled

Water-cooled

Water-cooled

Water-cooled

Water-cooled

MEASUREMENT CAPABILITY
Spectral range

0.19 - 20 µm

0.19 - 20 µm

0.19 - 20 µm

0.19 - 20 µm

0.19 - 20 μm

0.19 - 20 μm

Calibrated spectral range a

0.248 - 2.1 µm

0.248 - 2.1 µm

0.248 - 2.1 µm

0.248 - 2.1 µm

0.248 - 2.1 µm

0.248 - 2.1 µm

Noise equivalent power b

±3W

±3W

± 10 W± 10 W± 15 W± 15 W

Minimum average power c

100 W

100 W

300 W

300 W

500 W

500 W

Rise time (nominal)

7s

7s

9s

9s

15 s

15 s

Back reflections

10-15 %

< 4%

10 - 15 %< 4%

10 - 15 %2 - 4%

±5

± 5%

± 5%

± 5%

± 5%

± 5%

Repeatability

± 2%

± 2%

± 2%

± 2%

± 2%

± 2%

Linearity with power± 1.5 %

± 1.5 %

± 1.5 %

± 1.5 %

± 2%

± 2%

Linearity vs beam diameter

± 1%

± 1%

± 1%

± 1%

± 1%

± 1%

Linearity vs beam position e± 1.7 %

± 1.7 %

± 1.7 %

± 1.7 %

± 1.0 %

± 1.0 %

16 kW/cm2

16 kW/cm2

16 kW/cm2

16 kW/cm2---

Calibration uncertaintyd

BEAM PROFILING

4000 W

100 mm Ø

ENERGY DETECTORS

4000 W

EFFECTIVE APERTURE

POWER DETECTORSMAX AVERAGE POWER

HP100A-4KW-HE HP100A-4KWHE-TUBE

DAMAGE THRESHOLDS
Maximum average power density f
10 kW/cm2

10 kW/cm2

4 kW

4 kW/cm

4 kW/cm

---

---

---

5 kW

---

---

6.5 kW/cm2

6.5 kW/cm2

6.5 kW/cm2

6.5 kW/cm2

10 kW

---

---

3.5 kW/cm

3.5 kW/cm

2

3.5 kW/cm

3.5 kW/cm2

15 kW

---

---

---

---

1.5 kW/cm2

1.5 kW/cm2

100 mm Ø

70 mm Ø100 mm Ø

70 mm Ø125 x 125 mm

70 mm

HE

HE

HD

HD

HD

HD

Required cooling flow g(4 - 6) LPM< ± 1 LPM/min(4 - 6) LPM< ± 1 LPM/min(6 - 10) LPM< ± 1 LPM/min(6 - 10) LPM< ± 1 LPM/min(8 - 10) LPM< ± 1 LPM/min(8 - 10) LPM< ± 1 LPM/min

Temperature range

15 - 25 °C

15 - 25 °C

PHYSICAL CHARACTERISTICS
Effective aperture
Absorber (high damage threshold)

2

2

2

2

TERAHERTZ DETECTORS

500 W

Cooling water

15 - 25 °C

15 - 25 °C

15 - 25 °C< ± 3°C/min< ± 3°C/min< ± 3°C/min< ± 3°C/min< ± 3°C/min

Maximum water pressure (input) 413 kPa (60 psi)

413 kPa (60 psi)

413 kPa (60 psi)

413 kPa (60 psi)

413 kPa (60 psi)

413 kPa (60 psi)

Dimensions

127H x 127W x 74D mm

127H x 127W x 234D mm 127H x 127W x 70D mm

127H x 127W x 230D mm 153H x 153W x 70D mm

153H x 153W x 272D mm

Weight (head only)

1.8 kg

6.0 kg

3.3 kg

7.5 kg

5 kg

10 kg

DB15 & USB or
Bluetooth & USB

DB15 & USB or
Bluetooth & USB

DB15 & USB or
Bluetooth & USB

DB15 & USB or
Bluetooth & USB

DB15 & USB or
Bluetooth & USB

DB15 & USB or
Bluetooth & USB

STAND-S-443-C

2x STAND-S-443-C

STAND-S-443-C

2x STAND-S-443-C

2x STAND-S-443-C

3x STAND-S-443-C

ORDERING INFORMATION
Available output options
Compatible standProduct page

CUSTOM / OEM PRODUCTS

a. Calibrations at 2.1 to 2.5 μm and 10.6 μm are available on special request.
b. Nominal value, actual value depends on electrical noise in the measurement system.
c. For lower powers, call your Gentec-EO representative.
d. At 1064 nm and 1070 nm
e. For a beam size of 20 % of the aperture area, moved across 80 % of the aperture area.
f. At 1064 nm, 1.07-1.08 μm and 10.6 μm.
g. > 1 min. Contact Gentec-EO for deionized water cooling module option.
PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

DISPLAYS & PC INTERFACES

15 - 25 °C

Rate of temperature change < ± 3°C/min

.com

53

SUPER HP

Custom sizes and shapes, up to 120 000 W upon request

OUTPUT OPTIONSSMART DB15 CONNECTOR
Contains all the calibration data
USB PORT
Connects directly to a PC
Included in all HP models

COMPATIBLE DISPLAYS & PC INTERFACESKEY FEATURESMIRO ALTITUDE

MAESTRO

TUNER

Extension cables(4, 15, 20 or 25 m)

5 m USB cable(Included)

THE HIGHEST POWER HANDLING
Custom models handle up to 120 000 W ofcontinuous power
INFINITE CUSTOMIZATION CAPABILITIES
1. Choose YOUR size
2. Choose YOUR maximum power
3. We will customize one just for you!
COMPACT AND LIGHT WEIGHTLighter and more compact than any other high power
detector on the market, thanks to our unique design

UNO

ACCESSORIES	AVAILABLE WITH YAG AND CO2 CALIBRATIONSAll HP models can be calibrated at YAG and CO2
wavelengths with a calibration uncertainty of ± 5 %DIRECT USB CONNECTION TO A PC
Each head comes with both a DB15 connector (foruse with a Gentec-EO display device) and a USB2.0 output
for direct connection to a PC. Other connectors available
upon request

Stand with steel post
For 30 kW model

TRACK WATER PARAMETERS
Water flow and temperature are monitored in real timeand displayed continuously
HIGH POWER NIST-TRACEABLE CALIBRATIONWITH A 5 KW FIBER LASER

54

gentec-eo.com/laser-power-meter

Pelican carrying case

SUPER HP
Specifications

t

uc

ew

od
pr

N

HP100A-50KW-GD

CUSTOMIZATION CAPABILITIES

30 000 W

50 000 W

Up to 120 000 W

EFFECTIVE APERTURE

280 x 280 mm

100 mm Ø

Up to 500 x 500 mm

COOLING METHOD

Water-cooled

Water-cooled

Water-cooled

MEASUREMENT CAPABILITY
Spectral range

0.8 – 12 µm

0.19 - 20 µm

1.064 - 1.070 µm

1.030 - 1.080 µm

1.064 - 1.070 µm

Noise equivalent power b± 30 W± 200 W

Adapted to maximum power

Minimum average power c1000 W

1000 W

Adapted to maximum power

Rise time (nominal)

25 s

80 s

≤ 45 sBack reflections

10-15 %

< 2.5 %

Depends on the design

Calibration uncertainty d

± 5%

±5%

± 5%

Repeatability

± 2%

±2%

± 2%

Linearity with power

± 2%

±2%

± 2%

ENERGY DETECTORS

0.19 - 20 µm

Calibrated spectral range a

POWER DETECTORS

HP280A-30KW-HD
MAX AVERAGE POWER

BEAM PROFILINGDAMAGE THRESHOLDS
Maximum average power density e
10 kW

2.5 kW/cm2

25 kW/cm2

2.5 kW/cm2

30 kW

0.2 kW/cm

11 kW/cm

0.2 kW/cm2

280 x 280 mm

100 mm Ø

Square apertures up to 400 x 400 mm
Rectangular and round apertures also available

HD

GD

HD

Required cooling flow

0-30 kW: (15 - 18) LPM < ± 1 LPM/min f
0-10 kW: (12-15) LPM < ± 1 LPM/min f(18 - 25) LPM < ± 1 LPM/min

Temperature range

15 - 25 °C

15 - 25 °C

15 - 25 °C

Rate of temperature change< ± 3°C/min< ± 1°C/min< ± 3°C/min

Dimensions

314H x 324W x 89D mm

305H x 324W x 197D mm

Weight (head only)

20 kg

60 kg

DB15 & USB

DB15 & USB

DB15 & USB

STAND HP280A-30KW-HD

Ask

Ask

PHYSICAL CHARACTERISTICS
Effective aperture
Absorber (high damage threshold)22Compatible stand

Adapted to maximum power

DISPLAYS & PC INTERFACESORDERING INFORMATION
Available output options

TERAHERTZ DETECTORS

Cooling water

Product page

* These products are custom-built. Contact us with your requirements for a version tailored to your needs.

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

CUSTOM / OEM PRODUCTS

a. Calibrations at 0.248 to 2.5 µm and 10.6 µm are available on special request.
b. Nominal value, actual value depends on electrical noise in the measurement system.
c. For lower powers, call your Gentec-EO representative.
d. At 1064 nm and 1070 nm.
e. At 1064 nm, 1.07-1.08 µm and 10.6 µm.
f. > 1min

.com

55

HIGH-POWER PRONTO
1 W - 10 kW high power probes with touchscreen controls

USER INTERFACES (SSP MODE)

High damage thresholdbroadband absorber

Make a measurement in just a few seconds

5’ soft cable foreasy positionning

Ready

Countdown

Measurement complete!

The device waits
for a laser beam

Automatically starts when
exposed to a laser beam

The value is displayed
until the next measurement

Adjust the wavelength and calibration

Warns you when
the device is too hot*Set the brightness
and orientation

Press PLAY

Wavelength
Rugged casing
compact size

Calibration

Removable handle
with 1/4-20 mounting
hole for post

8-32 mounting
hole for post
Color LCD display withtouch screen controls
On/Offand main menu

USB connector for:- Battery charging- Data transfer to PC- Internet upgrades

KEY FEATURESACCESSORIESWIDE POWER RANGE
Very low noise level = wide power range with justone device
CONTINUOUS READINGS AT LOW POWERS
The PRONTO-500 includes a continuous power mode
(CWP) for measurements up to 40 W.
NO-WAIT MEASUREMENTS
5 seconds measurements allow for very short coolingtime (all models except PRONTO-3K)
EASY TO USE
The color LCD touchscreen allows for a friendly userinterface. You can make a measurement with just thetouch of a button!
DATA LOGGING
Save your data to the internal memory and thentransfer them to your PC over the USB connection.
LARGE APERTURE
55 mm Ø aperture to accommodate large beams
RUGGED• All-metal body• High damage thresholds
SERIAL COMMANDS
Serial commands are available to let you take fullcontrol of your PRONTO from your PC.

56

gentec-eo.com/laser-power-meterStand with steel post

Pelican carrying case

HIGH-POWER PRONTO
Specifications

PRONTO-3K

PRONTO-6K

PRONTO-10K

SSP Mode (Measures Power in 5 s)

500 W

3000 W

6000 W

10 000 W

CWP Mode (Measures Power continuously)

40 W

N/A

N/A

N/A

EFFECTIVE APERTURE

55 mm Ø55 mm Ø55 mm Ø55 mm ØCOOLING METHOD

Convection

Convection

Convection

Convection

MEASUREMENT CAPABILITY
Spectral range

0.19 - 20 µm

0.19 - 20 µm

0.19 - 20 µm

0.19 - 20 µm

Calibrated spectral range a

0.248 - 2.5 µm

0.248 - 2.5 µm

0.248 - 2.5 µm

0.248 - 2.5 µm

Noise equivalent power

0.1 W

5W

20 W

30 W

Exposure time

5 sb

10 s

5s

5s

Calibration uncertainty± 3 % (± 2.5 % in CWP mode)

± 5%

± 5%

± 5%

Number of readings before cooling c

100 W

25 (200 s)

0.5 kW

6 (72 s)

1 kW

6 (36 s)

1 kW

(Maximum exposure time before cooling)

200 W

12 (100 s)

1 kW

3 (36 s)

2 kW

3 (18 s)2 kW

5 (30 s)

300 W

8 (60 s)

1.5 kW

2 (24 s)

3 kW

2 (12 s)

5 kW

2 (12 s)

500 W

5 (40 s)

3 kW

1 (12 s)

6 kW

1 (6 s)

10 kW

1 (6 s)POWER DETECTORSPRONTO-500
MAX AVERAGE POWER

ENERGY DETECTORS

10 (60 s)

BEAM PROFILINGDAMAGE THRESHOLDS
Maximum average power density
25 kW/cm2

---

---

---

1064 nm, 500 W, CW

5 kW/cm2

7 kW/cm2------

1064 nm, 3000 W, CW---

5 kW/cm2

8 kW/cm2---

1064 nm, 6000 W, CW

---

---

7 kW/cm2

7 kW/cm2

1064 nm, 10 000 W, CW

---

---

-

5.5 kW/cm2

65 °C

65 °C

75 °C

75 °C

Maximum allowable casing temperature
GENERAL SPECIFICATIONS
Display type

Touchscreen color LCD

Touchscreen color LCD

Touchscreen color LCD

28.0 x 35.0 mm (128 x 160 pixels)

28.0 x 35.0 mm (128 x 160 pixels)

28.0 x 35.0 mm (128 x 160 pixels)

28.0 x 35.0 mm (128 x 160 pixels)

Data storage

50 000 pts

50 000 pts

50 000 pts

50 000 pts

Battery type

Rechargeable Li-ion

Rechargeable Li-ion

Rechargeable Li-ion

Rechargeable Li-ion

Battery life

17 hours or 4 200 measurements(with brightness set at 25 %)17 hours or 4 200 measurements(with brightness set at 25 %)17 hours or 4 200 measurements(with brightness set at 25 %)17 hours or 4 200 measurements(with brightness set at 25 %)Battery recharge via

USB port

USB port

USB port

USB port
55 mm ØPHYSICAL CHARACTERISTICS
Effective aperture

55 mm Ø55 mm Ø55 mm ØDimensions (sensor head)

88W x 88L x 32D mm

88W x 88L x 43D mm

88W x 88L x 36D mm

88W x 88L x 46D mm

Dimensions (monitor)

41W x 140L x 16D mm

41W x 140L x 16D mm

41W x 140L x 16D mm

41W x 140L x 16D mm

Weight930 g

1240 g1520 g2150 g

STAND-S-443

STAND-S-443

STAND-S-443

STAND-S-443

ORDERING INFORMATION
Compatible stand

DISPLAYS & PC INTERFACES

Touchscreen color LCD

Display size

TERAHERTZ DETECTORS1064 nm, 100 W, CW

Product page
CUSTOM / OEM PRODUCTS

a. For calibration at 10.6 μm, add C02-CAL-UP-2 to the orderb. Response time in CWP mode is 2 s.
c. Assuming an exposure time of 8 seconds and for 25°C starting temperature.

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

57

BD

Water-cooled beam dumps for high power lasers

REDUCE BACK-REFLECTIONSAll BD models can be fitted with a water-cooled absorbing TUBE to reducethe back-reflections below 4 %. The TUBE extension is backward-compatible
so it can be added to your current BD unit.

KEY FEATURESACCESSORIESEASY TO USE
Just plug the water-cooling and you’re done!
2 MODELS TO CHOOSE FROM
• 4 kW : BD-4KW-HE• 12 kW : BD-12KW-HD
VERY LARGE APERTURE
The round aperture of 100 mm in diameteraccommodates even the largest beams
HIGH DAMAGE THRESHOLDS
Up to 16 kW/cm2 (at 500 W)
ISOLATION TUBE IN OPTION
It is possible to add an isolation tube to
reduce back reflections

58

gentec-eo.com/laser-power-meterStand with steel post

Pelican carrying case

Water filter(Metric: 202984,
Imperial: 202990)

BD

Specifications

BD-12KW-HD

4000 W / 4500 W

12 000 W / 12 000 W

EFFECTIVE APERTURE

100 mm Ø

100 mm Ø

COOLING METHOD

Water-cooled

Water-cooled

500 W

10 kW/cm2

16 kW/cm2

4 kW

4 kW/cm2

---

5 kW

---

6.5 kW/cm2

10 kW

---

3.5 kW/cm2

Effective aperture

100 mm Ø

100 mm Ø

Absorber (high damage threshold)HE

HD

Required cooling flow

(4 - 6) LPM(6 - 10) LPM

Temperature range(15 - 25) °C(15 - 25) °C

Dimensions

127H x 127W x 74D mm

127H x 127W x 70D mm

Weight (head only)

1.8 kg

3.3 kg

STAND-S-443-C

STAND-S-443-C(CONTINUOUS / 2 MINUTES)POWER DETECTORS

BD-4KW-HE
MAX AVERAGE POWERDAMAGE THRESHOLDSENERGY DETECTORS

Maximum average power density a

PHYSICAL CHARACTERISTICS

Cooling water
BEAM PROFILINGORDERING INFORMATION
Compatible standProduct page
TERAHERTZ DETECTORS

a. At 1064 nm, 1.07-1.08 µm and 10.6 µm.

DISPLAYS & PC INTERFACESCUSTOM / OEM PRODUCTS

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

59

ABSORPTION CURVES
H, HD AND HE ABSORBERS
100%
90%
80%

Absorptance

70%
60%
50%
40%
30%
20%
10%
0%
0.1

1
Wavelength (µm)

10.6

VP ABSORBER
100%
90%
80%

Absorptance

70%
60%
50%
40%
30%
20%
10%
0%

0.25

2.5

Wavelength (µm)

60gentec-eo.com/laser-power-meter

25

ABSORPTION CURVESW ABSORBER
POWER DETECTORS

100%
90%
80%
70%
Absorptance

ENERGY DETECTORS

60%
50%
40%
30%
20%

0%

0.1

1

BEAM PROFILING

10%
10Wavelength (µm)

QED ABSORBERTERAHERTZ DETECTORS

100%
90%
80%

DISPLAYS & PC INTERFACES

Absorptance

70%
60%
50%
40%
30%
20%

CUSTOM / OEM PRODUCTS

10%
0%

0.1

1

10

Wavelength (µm)

PRODUCT GUIDE 2024

T 418.651.8003 | info@gentec-eo.com

.com

61

ABSORPTION CURVES

80

1000

70

875

60

750

50

625
OD2

40
30

500
Alone
375

20

With Attenuator (mW)

Without Attenuator (mW)

PH100-SI-HA maximum power

250
OD1

10
0
300

400

500

600

125
700

800

900

0
1100

1000

Wavelength (nm)

21

280

18

240
OD0.3

15

200

12

160

9

120
Alone

6

80
OD1

3
0
200

300

40

400

500

600

700

Wavelength (nm)62

gentec-eo.com/laser-power-meter

800

900

1000

0

With Attenuator OD1 or OD2 (mW)

Without Attenuator or With Attenuator OD0.3 (mW)

PH100-SIUV maximum powerABSORPTION CURVES
PH20-GE maximum power
600
OD2

525

60

450

50

375
300
OD1

30

225

20

150

10

75

900

1000

1100

1200

1300

1400

1500

1600

BEAM PROFILING

0800

ENERGY DETECTORS

Alone

40

With Attenuator (mW)70
Without Attenuator (mW)

POWER DETECTORS

80

0Wavelength (nm)800
700
Att ON
600

500

500

400

400

300

300

200

200

0
300

Att OFF

400

100

500

600

700

800

900

1000

CUSTOM / OEM PRODUCTS

600

Maximum Power with Attenuator (mW)

700

DISPLAYS & PC INTERFACES

Maximum Power without Attenuator (mW)

800

100

TERAHERTZ DETECTORSPRONTO-SI maximum power

0
1100

Wavelength (nm)PRODUCT GUIDE 2024

T 418.651.8003 | info@gentec-eo.com

.com

63

ENERGY DETECTOR
Low energy measurement

Available with

PE: PHOTODETECTORS• Available in 3 sizes:

3 mm Ø
5 mm Ø
10 mm Ø• 3 choices of absorber for different wavelength ranges:
SiliconGermanium
InGaAs• Extremely low noise: as low as 8 fJ

LOWEST NOISE LEVEL OF ALL ENERGY DETECTORS

Available with

QE-B: HIGH-SENSITIVITY PYROELECTRIC DETECTORS
Our pyroelectric energy detectors have very low noise levels combined with a large
bandwidth. They have everything you need to accurately measure extremely low
energy from the DUV to the FIR.
• 8 mm Ø aperture• 2 choices of absorber:

MT: Fast response and high sensitivity
BL: Flat spectral response• Broadband, from the DUV to the FIR• Very low noise: as low as 50 nJ

MEASURE LOW ENERGY AT ANY WAVELENGTH

MACH 6: MEASURE ALL PULSES UP TO 200 KHZ• High-speed digital joulemeter:
Measures EVERY PULSE at 200 kHz• Capture and store up to 4 million pulses at the maximum repetition rate
• Track missing pulses and pulses below threshold• Wide energy range: measure from pJ to mJ

200 kHz ENERGY METER

64

gentec-eo.com/laser-energy-meter

ENERGY DETECTORS
General use energy detectors

Available with

t

c
du

Pyroelectric energy meters cover a very wide range, going from nanojoules
to several tens of joules per pulse. Our standard absorber offers high damage
thresholds and a spectrally flat response, making this series of energy detectors a
versatile solution that can cover most of your energy measurement needs.

N

POWER DETECTORS

ew

o
pr

QE-MB

• Broadband absorber with high damage thresholds• Available in 7 sizes:

ENERGY DETECTORS

12 x 12 mm		95 mm Ø25 x 25 mm		NEW 145 mm Ø
50 x 50 mm		NEW 195 mm Ø
65 x 65 mm• Available with 2 cooling modules:
Convection (S)Heatsink (H)THE WIDEST RANGE OF LASER ENERGY MEASUREMENT
Available with

QE-MT: HIGH REPETITION RATESBEAM PROFILING

Designed for pulsed lasers with high repetition rates, these energy detectors
feature an improved temporal response to accurately measure pulse-to-pulse
energy at high repetition rates up to 10 kHz.
• Fast response, broadband absorber• Available in 3 sizes:

TERAHERTZ DETECTORS

12 x 12 mm
25 x 25 mm
50 x 50 mm• Available with 2 cooling modules:
Convection (S)Heatsink (H)UP TO 10 KHZ REPETITION RATE

pJ

nJ

µJ

mJ

J

Max rep. rate
1000 Hz200 000 Hz
1000 Hz10 000 Hz
4000 Hz
100 HzContact us1000 Hz
Single-shot
Single-shot
Single-shot

CUSTOM / OEM PRODUCTS

Spectral range
Depends on model
Depends on model
0.19 - 20 µm
0.19 - 20 µm*
0.19 - 20 µm*
0.19 - 20 µm*
0.19 - 20 µm*
Contact us
0.19 - 20 µm
0.19 - 20 µm
Contact us

PE
M6
QE-B
QE12, QE25QE50
QE65, QE95
QE145, QE195
IS50
PRONTOUP in SSE mode
Calorimeters

DISPLAYS & PC INTERFACESCOMPARISON TABLE - ENERGY MEASUREMENTkJ* QED models are represented by dashed area and have a limited spectral range: 0.3 - 2.1 µm

PRODUCT GUIDE 2024

T 418.651.8003 | info@gentec-eo.com

.com

65

ENERGY DETECTORSQED attenuators

The QED attenuators increase the maximum energy, energy density, average power and average power density that the QE series
detectors can handle. They are engineered to typically transmit 30-50 % (at 1064 nm) of the incident radiation to the detector in anear Lambertian pattern (very wide diffusion pattern). Their slide-in casing make them easy to install and remove and they are held
securely in place with the use of simple set screws. Since they become part of the detector, it is important to understand how they
will affect the calibration.

CALIBRATION OPTIONS
Depending on how you plan to use a QE detector and QED attenuator, different purchasing and calibration options are available.

QE detector with QED attenuator included
Product name contains “-QED”
Ex: QE25LP-S-MB-QED-D0
This product is calibrated with the QED attenuator in place. You
may remove the attenuator, but your measurements will not becalibrated with this configuration.

QE detector and QED attenuator purchased separately
Product name does not contain “-QED”
Ex: QE25LP-S-MB-D0 and QED-25
Three calibration options are available when you purchase the QE
detector and the QED attenuator separately.

OR

FULL CALIBRATION
The detector is fully
calibrated both with andwithout attenuator. Thisconfiguration comes with a
DB15 adaptor.
• QED-CAL-3

PARTIAL CALIBRATION
The detector is fully
calibrated without
attenuator, and is calibrated
at a single wavelength withthe attenuator.
• QED-CAL-1

Fully calibrated

Fully calibrated

Fully calibrated

Fully calibrated when using
the DB15 adaptor

Calibrated at 1064 nm only

Not calibrated

Detector alone
Detector with attenuator

NO EXTRA CALIBRATIONThe QE detector is fully
calibrated without
attenuator only. You may
add the attenuator, but
your measurements will
not be calibrated with thisconfiguration.

SPECIFICATIONSPHYSICAL CHARACTERISTICS

66

QED-12

QED-25

QED-50

QED-65

QED-95Spectral range

266 - 2100 nm

266 - 2100 nm

266 - 2100 nm

266 - 2100 nm

266 - 2100 nm

Calibrated spectral range

532 - 2100 nm

308 - 2100 nm

308 - 2100 nm

308 - 2100 nm

308 - 2100 nm

Effective aperture

9 x 9 mm

22 x 22 mm

47 x 47 mm

62 x 62 mm

90 mm ØDimensions

30.5H x 41W x 12.5D mm

44H x 55W x 12.5D mm

69H x 80W x 12.5D mm

85H x 97W x 12.5D mm

115H x 127W x 12.5D mm

For use with

QE12

QE25

QE50

QE65

QE95

gentec-eo.com/laser-energy-meter

ENERGY DETECTORSHigh energy detectors

IS50: ENERGY METER FOR HIGH AVERAGE POWERtuc

Custom-built to your specifications, contact us with your laser measurement
needs

N

• Designed for high energy measurements at high repetition rates• Can handle up to 1000 W average power• Our proprietary coating offers damage thresholds that are orders ofmagnitude higher than any other “white” coating on the market.

POWER DETECTORS

ew

od
pr

I DEAL FOR IPL SOURCES: UP TO 350 J
ENERGY DETECTORS

Available with

THERMOPILES IN SINGLE-SHOT ENERGY MODE
MEASURE ENERGY WITH A POWER DETECTORS EE “ENERGY MODE” IN THE POWER DETECTOR SPECIFICATIONSBEAM PROFILING

The single-shot energy mode, available with all our thermal power detectors,
allows you to measure the energy of single pulses or pulse trains.

PRONTO-500-IPL
TERAHERTZ DETECTORS• Compact energy meter for up to 350 J• 55 mm Ø aperture• Color touchscreen display• Rugged device: all-metal body and protective windowI DEAL FOR IPL SOURCES: UP TO 350 J

We work with a wide range of materials from surface coatings to the mostrobust volume absorbers to provide the best solution for your specificapplication.
• Outstanding signal-to-noise ratios• Vacuum compatibility• Attention to detail and workmanshipWith over 50 years of experience in thermal-based energy measurement,
Gentec-EO is the ideal choice for all your high energy measurement needs.
* Maximum measurable energy depends on pulse width and wavelength.
PRODUCT GUIDE 2024

T 418.651.8003 | info@gentec-eo.com.com

CUSTOM / OEM PRODUCTS• High sensitivity

DISPLAYS & PC INTERFACESCUSTOM CALORIMETERS

67

PE

8 fJ - 150 nJ, our lowest energy measurements

OUTPUT OPTIONSSMART INTERFACE
Containing all the calibration data
ANALOG OUTPUT
When used with APM (D) analog power supply
ALL-IN-ONE-METER
Connects directly to a PC
Three models available:
USB output (-INT)
RS-232 output (-IDR)
USB with external trigger (-INE)

COMPATIBLE DISPLAYS & PC INTERFACESKEY FEATURESMIRO ALTITUDE

MAESTRO

M-LINK

S-LINK

U-LINKACCESSORIESVERY LOW NOISE LEVELTake measurements with a noise level as low as 8 fJ
(model PE3B-Si only) with the M-LINK, MAESTRO andS-LINK
3 SENSORS AVAILABLE• PE-B-SI family: 3 and 10 mm Ø silicon sensors
for 0.21 to 1.08 μm

Stand with delrin post

Fiber adaptors & connectors(FC, ST or SMA)

Pelican carrying case

Isolation tube

• PE5B-GE: 5 mm Ø, germanium sensor
for 0.8 to 1.65 μm• PE3B-IN: 3 mm Ø, InGaAs sensor
for 0.9 to 1.7 μm

This product cannot be used with DB15 extension cables

68

gentec-eo.com/laser-energy-meter

APM (D) analog power
supplyPESpecifications

PE10B-SI-D0

PE5B-GE-D0

PE3B-IN-D0

MAX MEASURABLE ENERGY*

24 pJ

81 nJ

2.4 nJ

245 pJEFFECTIVE APERTURE

3 mm Ø

10 mm Ø5 mm Ø

3 mm Ø

210 - 1080 nm

210 - 1080 nm

800 - 1650 nm

900 - 1700 nm

With M-LINK

22 pJ at 634 nm

75 nJ at 634 nm

2.2 nJ at 1310 nm

223 pJ at 1310 nm

With S-LINK

24 pJ at 634 nm

81 nJ at 634 nm

2.4 nJ at 1310 nm

245 pJ at 1310 nm

With MAESTRO

20 pJ at 634 nm

69 nJ at 634 nm

2.0 nJ at 1310 nm

200 pJ at 1310 nm

With INTEGRA

24 pJ at 634 nm

81 nJ at 634 nm

2.4 nJ at 1310 nm

245 pJ at 1310 nm

8 fJ at 634 nm

1.5 pJ at 634 nm

1 pJ at 1310 nm

30 fJ at 1310 nm

Rise time (0-100 %)

15 µs

30 µs

25 µs

12 µs

Max repetition rate

1000 Hz

1000 Hz

1000 Hz

1000 HzMax pulse width

10 µs

10 µs

10 µs

10 µs

± 18 % (210 - 229 nm)± 5 % (800 - 1049 nm)± 4% d

± 8.0 % (230 - 254 nm)± 3.5 % (1050 - 1559 nm)± 6.5 % (255 - 399 nm)± 7 % (1560 - 1629 nm)± 2.5 % (400 - 899 nm)± 10 % (1630 - 1650 nm)

POWER DETECTORS

PE3B-SI-D0

MEASUREMENT CAPABILITY
Calibrated spectral range
Maximum measurable energy*

± 4%

c

BEAM PROFILING

Calibration uncertaintyb

ENERGY DETECTORS

Noise equivalent energya± 4.0 % (900 - 1009 nm)± 7.5 % (1010 - 1080 nm)

DAMAGE THRESHOLDS
N/A

5 µJ/cm2

5 µJ/cm2

N/A

Max average power density

N/A

65 mW/cm2 at 532 nm

320 mW/cm2 at 1064 nm

N/A

Effective aperture

3 mm Ø

10 mm Ø5 mm Ø

3 mm Ø

Distance to sensor face

13.7 mm

13.7 mm10.5 mm

N/A

SensorUV-silicon

UV-silicon

Germanium

InGaAsDimensions

38.1Ø x 27.4D mm

38.1Ø x 27.4D mm

38.1Ø x 27.4D mm

38.1Ø x 27.4D mm

Weight

91 g

91 g

91 g

91 g

Available output options

DB15, USB or RS-232

DB15, USB or RS-232

DB15, USB or RS-232

DB15, USB or RS-232

Compatible stand

STAND-D-233 or STAND-D-233-M STAND-D-233 or STAND-D-233-M STAND-D-233 or STAND-D-233-M STAND-D-233 or STAND-D-233-M

TERAHERTZ DETECTORS

Max energy density
PHYSICAL CHARACTERISTICS

Product page

CUSTOM / OEM PRODUCTS* See curves (p. 102-103) for maximum energy at other wavelengths
a. Nominal value. Depends on environmental electromagnetic interference and wavelength.
b. With Gentec-EO display or PC interface.
c. This detector is NIST Traceable at the calibration wavelength of 634 nm. Typical values are used at other wavelengths.
d. This detector is NIST Traceable at the calibration wavelength of 1310 nm. Typical values are used at other wavelengths.

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

DISPLAYS & PC INTERFACESORDERING INFORMATION

.com

69

QE-B

50 nJ - 3.6 mJ, ultra-low energy measurements

OUTPUT OPTIONSSMART INTERFACE
Containing all the calibration data
ANALOG OUTPUT
When used with APM (D) analog power supply
ALL-IN-ONE-METER
Connects directly to a PC
Three models available:
USB output (-INT)
RS-232 output (-IDR)
USB with external trigger (-INE)

COMPATIBLE DISPLAYS & PC INTERFACESKEY FEATURESMIRO ALTITUDE

MAESTRO

M-LINK

S-LINK

U-LINKACCESSORIESVERY LOW NOISE LEVELS
Noise levels of a photodetector, but with the highenergies of a pyroelectric:• 50 nJ with the MT coating• 100 nJ with the BL coating
2 COATINGS AVAILABLE

Stand with delrin post

Fiber adaptors & connectors(FC, ST or SMA)

Pelican carrying case

Isolation tube

• BL: B lack coating, sensitivity of 900 V/J,
readings up to 400 Hz• MT: M etallic coating, sensitivity of 2400 V/J,
readings up to 1000 Hz

This product cannot be used with DB15 extension cables

70

gentec-eo.com/laser-energy-meter

APM (D) analog power supply

QE-B
Specifications

QE8SP-B-MT

MAX MEASURABLE ENERGY

3.6 mJ

1.3 mJMAX REPETITION FREQUENCY

400 Hz1000 Hz

EFFECTIVE APERTURE

7.8 X 7.8 mm

7.8 X 7.8 mm

Spectral range

0.19 - 20 µm

0.19 - 20 µm

Calibrated spectral range a

0.248 - 2.1 µm

0.248 - 2.1 µm

With U-LINK

3.6 mJ

1.3 mJ

With S-LINK

2.9 mJ

1.1 mJ

With MAESTRO

2.5 mJ

0.93 mJ

With U-LINK

150 nJ

80 nJ

With S-LINK

100 nJ

50 nJ

POWER DETECTORSQE8SP-B-BL

MEASUREMENT CAPABILITY
ENERGY DETECTORS

Max measurable energy

Noise equivalent energy

150 nJ

80 nJ

400 Hz1000 HzMax pulse width

10 µs

10 µs

Risetime (0-100 %)

30 µs

30 µs

Calibration uncertainty± 4.0 %

± 4.0 %

Repeatability< 0.5 %

< 0.5 %

0.5 W

0.5 W

1 W/cm2

1 W/cm2

50 mJ/cm2

50 mJ/cm2

Effective aperture

7.8 x 7.8 mm

7.8 x 7.8 mm

AbsorberOrganic black

Metallic

Dimensions

38.1 Ø X 27.4D mm

38.1 Ø X 27.4D mm

Weight91 g

91 g

Available output options

DB15, USB or RS-232

DB15, USB or RS-232

Compatible stand

STAND-D-233 or STAND-D-233-M

STAND-D-233 or STAND-D-233-M

BEAM PROFILING

With MAESTRO
Max repetition frequency

DAMAGE THRESHOLDSTERAHERTZ DETECTORS

Maximum average power
Maximum everage power density
1064 nm, 7 ns, 10 Hz
Maximum energy density
1064 nm, 7 ns, 10 Hz
PHYSICAL CHARACTERISTICSDISPLAYS & PC INTERFACESORDERING INFORMATION

Product page

CUSTOM / OEM PRODUCTS

a. Calibration at 2.1 to 2.5 µm is available on special request.

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

71

MACH 6200 kHz energy meter

COMPATIBLE DISPLAYS & PC INTERFACES

MACH 6KEY FEATURES

APM

ACCESSORIESUP TO 200 kHz PULSE-TO-PULSE
Measure EVERY pulse, with no sampling, at high rep rates,
up to 200 kHz
CAPTURE AND STORE UP TO 4 MILLION PULSESStore 40 seconds of data at 100 kHz
	TRACK MISSING PULSES AND PULSES BELOWTHRESHOLD
Know how many pulses were missed or that didn’t makethe energy threshold with this unique pulse featureStand with delrin post

Additional 9V power supply

USB cable

APM analog power supply(requires adaptor when usedwith M6 heads)

M6-UV-QED
Relative measurements in UV

Pelican carrying case

SEVERAL HEADS TO CHOOSE FROMSilicon, InGaAs and pyroelectric heads for a broad
wavelength and energy range
ANALOG MODULE AVAILABLE
Use our fast M6 detectors with the APM and an
oscilloscope for fast analog energy measurements
FULL-SPEED USB 2.0 CONNECTION
Ensures high data rate transfer and fast operation
USER-FRIENDLY SOFTWARE WITH MANYDIAGNOSTIC FEATURES• Live mode, strip chart, histogram and statistics displays• FFT display of pulse energy data for temporal
diagnostics• Life test mode to automate laser testing72

gentec-eo.com/laser-energy-meter

MACH 6
Specifications

M6-6-SI-L

M6-6-IN

M6-6-IN-L

M6-6-PY

M6-12.5-PY

MAX ENERGY

200 µJ

2 µJ

200 µJ

2 µJ

2 mJ

20 mJ

MAX AVERAGE POWER

5W

5W

5W

5W

5W

25 W

MAX REP RATE

200 000 Hz

200 000 Hz

200 000 Hz

200 000 Hz

200 000 Hz

200 000 Hz

EFFECTIVE APERTURE

6 mm Ø

6 mm Ø

6 mm Ø

6 mm Ø

6 mm Ø

12.5 mm Ø

0.35 - 1.1 µm

0.35 - 1.1 µm

0.9 - 1.6 µm

0.9 - 1.6 µm

0.35 - 2.5 µm

0.35 - 2.5 µm

200 µJ

2 µJ

200 µJ

2 µJ

2 mJ

20 mJ

Noise equivalent energy

2 nJ

20 pJ

2 nJ

20 pJ

0.2 µJ

0.2 µJ

Rise time (0-100 %)150 ns

150 ns

150 ns

150 ns

150 ns

150 nsMax repetition rate

200 000 Hz

200 000 Hz

200 000 Hz

200 000 Hz

200 000 Hz

200 000 Hz

Max pulse width

100 ns

100 ns

100 ns

100 ns

100 ns

100 nsCalibration uncertainty

± 4%

± 4%

± 4%

± 4%

± 4%

± 4%

Repeatability

± 1%

± 1%

± 1%

± 1%

± 1%

± 1%

Max average power (60 seconds)

5W

5W

5W

5W

5W

25 W

Max energy (at 1064 nm)

200 µJ

2 µJ

200 µJ

2 µJ

20 mJ

20 mJ

Effective aperture

6 mm Ø

6 mm Ø

6 mm Ø

6 mm Ø

6 mm Ø

12.5 mm Ø

SensorSiliconSiliconInGaAsInGaAsPyroelectric

Pyroelectric

Dimensions

38.1 Ø x 58.4D mm

38.1 Ø x 58.4D mm

38.1 Ø x 58.4D mm

38.1 Ø x 58.4D mm

38.1 Ø x 58.4D mm

76H x 111W x 76D

Weight (head only)

150 g

150 g

150 g

150 g

150 g

N/A

STAND-D-233

STAND-D-233

POWER DETECTORS

M6-6-SIa

MEASUREMENT CAPABILITY
Max measurable energya

ENERGY DETECTORS

Spectral range

BEAM PROFILINGDAMAGE THRESHOLDSPHYSICAL CHARACTERISTICSTERAHERTZ DETECTORSCOMPATIBLE METERS
PC-based

Mach 6: see detailed specifications on next page

Analog power supply

APM: see detailed specifications on next page (requires adaptor when used with M6 heads)

ORDERING INFORMATION
Compatible stand

STAND-D-233

STAND-D-233

STAND-D-233

STAND-D-233

Product pageDISPLAYS & PC INTERFACES

a. Maximum pulse energy reading will vary with wavelength and probe voltage responsivity (Rv). For more details, please read Application Note 121D-201932 and contact us at info@gentec-eo.com

CUSTOM / OEM PRODUCTS

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

73

MACH 6
Specifications

MACH 6 JOULEMETER
Measure every pulse at up to 200 kHz with MACH 6. Measure with 12-bit
digital accuracy and capture up to 4 million pulses in real time. Our MACH 6
joulemeter is the only instrument in the world that performs at this speed, andwith this precision. It is designed to support our full complement of fast energy
probes that include silicon, InGaAs and pyroelectric detectors. Measure from pJ
to mJ and from 0.35 to 2.5 µm. Using the M6-Si detector and the M6-UV-QED
accessory, you can make relative measurements at 266 nm.

SPECIFICATIONS & FEATURES

MACH 6(rear view)

APM analog
power supply(front view)

MACH 6Compatible detector heads M6APM
M6 (with adaptor: P/N C201949), UM-B,
QE8, THZ9D and PE detectors

Maximum repetition rate

200 000 Hz

Limited by oscilloscope and detector

Analog output

0-3V± 4.88 V, BNC

External trigger (TTL)

Optically coupled

NoneInternal trigger

2 - 20 %None

Trigger delay

38 - 3825 ns (user-settable)

None

Digital output

USB 2.0

NonePower supply

External, 100/240 VAC 50 - 60 Hz External, 100/240 VAC 50 - 60 Hz,
and 9 V battery (both included)

Product number

202090201848INSTRUMENT CONTROL SCREEN
Our powerful LabVIEW application software includes many unique control
and diagnostic features. The instrument control screen, shown on the left,
is used to set up the operation of the MACH 6, including range, trigger,
wavelength, and more. In addition, it is used to set a pulse batch size and toarm the instrument, which starts the data collection. It also gives you accessto features like “Autoset”, “Call Live Mode”, “Run Life Test”, “Save Instrument
Setup” and the like. These features can be accessed by clicking directly on thefeature or pushing the associated function key.

LIVE MODE DISPLAY
The live mode can be accessed from the instrument control screen, or bysimply pressing the “F4” function key on your PC. This mode of operation isintended to act like your typical slower digital joulemeter, as it samples thelaser pulses at a 10 Hz rate. It provides you with an energy strip chart, live
energy reading, statistics and repetition rate. It is a very useful mode whensetting up the Mach 6 with your laser. You can select “Auto Set”, where theinstrument runs through the ranges and trigger levels until it finds the correct
range, or set them manually. When setup is complete, you will exit this screen
and return to the “Control” screen where you will select a batch size, arm the
instrument and start taking pulse energy data.

74

gentec-eo.com/laser-energy-meter

MACH 6
Specifications

STRIP CHART

ENERGY DETECTORSHISTOGRAM

BEAM PROFILINGInterested in viewing the statistical distribution of the pulse energy data
set? The Histogram screen does this for you and fits a “best Gaussian curve”
to the data. It displays complete statistical calculations along the bottom of
the graph, along with pulse frequency. Just below the statistics, you will findinstrument controls, like range and trigger. You are also given the ability toadjust trigger delay and hold off as needed.

The statistics display offers a very complete set of useful energy readings andcalculated statistics. These include: minimum, maximum, average (mean),standard deviation, spread and variation. Some other very handy features
include: windows displaying, average frequency, pulse jitter, pulses below trigger
and pulses below threshold (a level set by you). In the screen on the left, you
can see that there were 12 pulses below an energy threshold of 3 µW, and 147
missing pulses (or pulses below trigger).

DISPLAYS & PC INTERFACES

LIFE TEST MODE

CUSTOM / OEM PRODUCTS

Would you like to run a life test on your high repetition rate pulsed laser? Howabout a periodic test vs. an environmental change like temperature? We haveincluded a great feature to accomplish this. In the life test screen, we give you
the ability to select the statistics you want, a start time and date, a stop timeand date, the number of pulses and a test interval. You simply identify a file, aplace to put the data, and then click on start and walk away! When you comeback, you have a data set that tracked the performance of your laser over time,
temperature, shock, vibration or anything you chose.

T 418.651.8003 | info@gentec-eo.com

TERAHERTZ DETECTORSSTATISTICS

PRODUCT GUIDE 2024

POWER DETECTORSThe strip chart display provides a quick graphical look at the pulse data batch
just collected. The data can be displayed in full scale or in auto scale mode.
You can also zoom-in on a portion of the data, like shown in the screen on
the left. An ND0.3 filter has been dropped through the beam and you can see
the effects on the pulse data collected. You can fit trend and min/max lines
to the data. Just below the chart, you will find a complete set of statistics forthe batch. At the top of this screen you will see tabs that will take you to theHistogram, Statistics, and FFT (Fast Fourier Transform) displays.

.com

75

QE12-MB
12 x 12 mm, 0.7 µJ - 3.9 J

OUTPUT OPTIONSSMART INTERFACE
Containing all the calibration data
ALL-IN-ONE-METER
Connects directly to a PC
Three models available:
USB output (-INT)
RS-232 output (-IDR)
USB with external trigger (-INE)

COMPATIBLE DISPLAYS & PC INTERFACESKEY FEATURESMIRO ALTITUDE

MAESTRO

M-LINK

S-LINK

U-LINKACCESSORIESMODULAR CONCEPT
Increase the power capability of your detector:
2 different cooling modules
LOW NOISE LEVELQED ATTENUATOR AVAILABLE• Measure up to 5X higher energies

Stand with delrin post

• Available with optional calibration,
all wavelengths between 532 & 1064 nm,
or single wavelength
HIGH REPETITION RATE OPTIONS• QE12LP: 300 Hz• QE12HR: 1000 Hz
TEST TARGET INCLUDED

76

gentec-eo.com/laser-energy-meter

Pelican carrying case

DB15 to BNC adaptor

QED-12 attenuator

QE12-MB
Specifications

QE12LP-H-MB

QE12LP-H-MBQEDQE12HR-H-MB

QE12HR-H-MBQED

3.9 J

3.9 J

3.9 J

3.9 J

0.85 J

3.9 J

300 Hz300 Hz300 Hz300 Hz

1 kHz

1 kHz

12 x 12 mm

9 x 9 mm

12 x 12 mm

9 x 9 mm

12 x 12 mm

9 x 9 mm

MEASUREMENT CAPABILITY
Spectral range

0.19 - 20 µm

0.3 - 2.1 µm

0.19 - 20 µm

0.3 - 2.1 µm

0.19 - 20 µm

0.3 - 2.1 µm

Calibrated spectral range

0.248 - 2.1 µm

0.532 - 2.1 µm

0.248 - 2.1 µm

0.532 - 2.1 µm

0.248 - 2.1 µm

0.532 - 2.1 µm

1064 nm, 7 ns

0.85 J

3.9 J

0.85 J

3.9 J

0.85 J

3.9 J

266 nm, 7 ns

0.70 J

0.81 J

0.70 J

0.81 J

0.70 J

0.81 JMAX MEASURABLE ENERGY a
MAX REPETITION FREQUENCYEFFECTIVE APERTURE

cb

ENERGY DETECTORSQE12LP-S-MBQED

POWER DETECTORSQE12LP-S-MB

Maximum measurable energy a

Noise equivalent energy

1.4 µJ

0.7 µJ

1.4 µJ

1.4 µJ

2.8 µJ

300 Hz300 Hz300 Hz300 Hz

1 kHz

1 kHz

Maximum pulse width (typical) e

400 µs400 µs400 µs400 µs

40 µs

40 µs

Calibration uncertainty f

± 3%

± 3%

± 3%

± 3%

± 3%

± 3%

Repeatability< 0.5 %

< 0.5 %

< 0.5 %

< 0.5 %

< 0.5 %

< 0.5 %

3W

7.5 W

5W

12.5 W

5W

12.5 W1064 nm, 7 ns, single shot

0.6 J/cm2

16 J/cm2

0.6 J/cm2

16 J/cm2

0.6 J/cm2

16 J/cm2

1064 nm, 7 ns, 10 Hz

0.6 J/cm2

8 J/cm2

0.6 J/cm2

8 J/cm2

0.6 J/cm2

8 J/cm2

532 nm, 7 ns, 10 Hz

2

0.6 J/cm

2

6 J/cm2

0.6 J/cm

2

6 J/cm2

0.6 J/cm

6 J/cm2

266 nm, 7 ns, 10 Hz

2

0.5 J/cm

2

1 J/cm2

0.5 J/cm

2

1 J/cm2

0.5 J/cm

1 J/cm2

600 W/cm2

10 W/cm2 j

600 W/cm2

10 W/cm2

600 W/cm2

12 x 12 mm

9 x 9 mm

12 x 12 mm

9 x 9 mm

12 x 12 mm

9 x 9 mm

Absorber

MB

QED

MB

QED

MB

QED

Dimensions

36H x 36W x 14D mm

39H x 41W x 19D mm

36H x 36W x 33D mm

39H x 41W x 38D mm

36H x 36W x 33D mm

39H x 41W x 38D mm

Weight

87 g

87 g

117 g

117 g

117 g

117 g

Available output options

DB15, USB or RS-232

DB15, USB or RS-232

DB15, USB or RS-232

DB15, USB or RS-232

DB15, USB or RS-232

DB15, USB or RS-232

Compatible stand

STAND-D-233

STAND-D-233

STAND-D-233

STAND-D-233

STAND-D-233

STAND-D-233

DAMAGE THRESHOLDS
Maximum average power

BEAM PROFILING

0.7 µJ

Max repetition frequency bdMaximum energy density

PHYSICAL CHARACTERISTICS
Effective aperture (with attenuator)

TERAHERTZ DETECTORS

Maximum average power density g 10 W/cm2

DISPLAYS & PC INTERFACESORDERING INFORMATION

Product page

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com.com

CUSTOM / OEM PRODUCTS

a. Not exceeding maximum average power. Increasing pulse width increases the maximum measurable energy. The maximum measurable energy depends on the display or PC interface used.
If your laser is close to the maximum, contact us to check your specifications.
b. With the IDR version, measured values are sampled when the repetition rate is > 200 Hz.
c. Calibration at 2.1 to 2.5 µm is available on special request.
d. Nominal value, actual value depends on electrical noise in the measurement system.
e. Also available on special order: ELP (extra-long pulse) version.
f. Excludes non-linearities.
g. At maximum power.

77

QE25-MB
25 x 25 mm, 2 µJ - 23 J

OUTPUT OPTIONSSMART INTERFACE
Containing all the calibration data
ALL-IN-ONE-METER
Connects directly to a PC
Three models available:
USB output (-INT)
RS-232 output (-IDR)
USB with external trigger (-INE)

COMPATIBLE DISPLAYS & PC INTERFACESKEY FEATURESMIRO ALTITUDE

MAESTRO

M-LINK

S-LINK

U-LINKACCESSORIESMODULAR CONCEPT
Increase the power capability of your detector:
2 different cooling modules
LOW NOISE LEVELQED ATTENUATOR AVAILABLE• Measure up to 5X higher energies

Stand with delrin post

• Available with optional calibration,
all wavelengths between 532 & 1064 nm,
or single wavelength
HIGH REPETITION RATE OPTIONS• QE25LP: 300 Hz• QE25HR: 1000 Hz
TEST TARGET INCLUDED

78

gentec-eo.com/laser-energy-meter

Pelican carrying case

DB15 to BNC adaptor

QED-25 attenuator

QE25-MB
Specifications

QE25LP-S-MBQEDQE25LP-H-MB

QE25LP-H-MBQEDQE25HR-H-MB

QE25HR-H-MBQEDMAX MEASURABLE ENERGY a

3.8 J

23 J

3.8 J

23 J

3.8 J

23 J

MAX REPETITION FREQUENCY b

300 Hz300 Hz300 Hz300 Hz

1 kHz

1 kHz

EFFECTIVE APERTURE

25 x 25 mm

22 x 22 mm

25 x 25 mm

22 x 22 mm

25 x 25 mm

22 x 22 mm

0.19 - 20 µm

0.3 - 2.1 µm

0.19 - 20 µm

0.3 - 2.1 µm

0.19 - 20 µm

0.3 - 2.1 µm

0.248 - 2.1 µm

0.308 - 2.1 µm

0.248 - 2.1 µm

0.308 - 2.1 µm

0.248 - 2.1 µm

0.308 - 2.1 µm

1064 nm, 7 ns

3.8 J

23 J

3.8 J

23 J

3.8 J

23 J

266 nm, 7 ns

3.1 J

4.8 J

3.1 J

4.8 J

3.1 J

4.8 J

POWER DETECTORSQE25LP-S-MB

MEASUREMENT CAPABILITY
Calibrated spectral rangec

ENERGY DETECTORS

Spectral range
Maximum measurable energy a

Noise equivalent energy

8 µJ

4 µJ

8 µJ

10 µJ

20 µJ

300 Hz300 Hz300 Hz300 Hz

1 kHz

1 kHz

Maximum pulse width (typical) e

400 µs400 µs400 µs400 µs

40 µs

40 µs

Calibration uncertainty f

± 3%

± 3%

± 3%

± 3%

± 3%

± 3%

Repeatability< 0.5 %

< 0.5 %

< 0.5 %

< 0.5 %

< 0.5 %

< 0.5 %

5W

15 W

10 W

30 W

10 W

30 W

1064 nm, 7 ns, single shot

0.6 J/cm2

16 J/cm2

0.6 J/cm2

16 J/cm2

0.6 J/cm2

16 J/cm2

1064 nm, 7 ns, 10 Hz

0.6 J/cm2

8 J/cm2

0.6 J/cm2

8 J/cm2

0.6 J/cm2

8 J/cm2

532 nm, 7 ns, 10 Hz

0.6 J/cm2

6 J/cm2

0.6 J/cm2

6 J/cm2

0.6 J/cm2

6 J/cm2

266 nm, 7 ns, 10 Hz

0.5 J/cm

1 J/cm

0.5 J/cm

1 J/cm

0.5 J/cm

1 J/cm2

600 W/cm2

10 W/cm2

600 W/cm2

10 W/cm2

600 W/cm2

25 X 25 mm

22 X 22 mm

25 X 25 mm

22 X 22 mm

25 X 25 mm

22 X 22 mm

Absorber

MB

QED

MB

QED

MB

QED

Dimensions

50H x 50W x 14D mm

53H x 55W x 19D mm

50H x 50W x 53D mm

53H x 55W x 58D mm

50H x 50W x 53D mm

53H x 55W x 58D mm

Weight

120 g

120 g

193 g

193 g

193 g

193 g

Available output options

DB15, USB or RS-232

DB15, USB or RS-232

DB15, USB or RS-232

DB15, USB or RS-232

DB15, USB or RS-232

DB15, USB or RS-232

Compatible stand

STAND-D-233

STAND-D-233

STAND-D-233

STAND-D-233

STAND-D-233

STAND-D-233

DAMAGE THRESHOLDS
Maximum average power

BEAM PROFILING

4 µJ

Max repetition frequency bdMaximum energy density

Maximum average power density g 10 W/cm2
PHYSICAL CHARACTERISTICS
Effective aperture

2

2

2

2

DISPLAYS & PC INTERFACESORDERING INFORMATION

Product page

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com.com

CUSTOM / OEM PRODUCTS

a. Not exceeding maximum average power. Increasing pulse width increases the maximum measurable energy. The maximum measurable energy depends on the display or PC interface used.
If your laser is close to the maximum, contact us to check your specifications.
b. With the IDR version, measured values are sampled when the repetition rate is > 200 Hz.
c. Calibration at 2.1 to 2.5 µm is available on special request.
d. Nominal value, actual value depends on electrical noise in the measurement system.
e. Also available on special order: ELP (extra-long pulse) version
f. Excludes non-linearities.
g. At maximum power.

PRODUCT GUIDE 2024

TERAHERTZ DETECTORS279

QE50-MB
50 x 50 mm, 10 µJ - 85 J

OUTPUT OPTIONSSMART INTERFACE
Containing all the calibration data
ALL-IN-ONE-METER
Connects directly to a PC
Three models available:
USB output (-INT)
RS-232 output (-IDR)
USB with external trigger (-INE)

COMPATIBLE DISPLAYS & PC INTERFACESKEY FEATURESMIRO ALTITUDE

MAESTRO

M-LINK

S-LINK

U-LINKACCESSORIESMODULAR CONCEPT
Increase the power capability of your detector:
2 different cooling modules
LOW NOISE LEVELQED ATTENUATOR AVAILABLE

Stand with delrin post

• Measure up to 5X higher energies• Available with optional calibration,
all wavelengths between 532 & 1064 nm,
or single wavelength
TEST TARGET INCLUDED
Pelican carrying case80

gentec-eo.com/laser-energy-meter

DB15 to BNC adaptor

QED-50 attenuator

QE50-MB
Specifications

QE50LP-H-MB

QE50LP-H-MB-QED

85 J

15 J

85 J

MAX REPETITION FREQUENCY

200 Hz200 Hz200 Hz200 Hz

EFFECTIVE APERTURE

50 x 50 mm

47 x 47 mm

50 x 50 mm

47 x 47 mm

MEASUREMENT CAPABILITY
Spectral range

0.19 - 20 µm

0.3 - 2.1 µm

0.19 - 20 µm

0.3 - 2.1 µm

Calibrated spectral range b

0.248 - 2.1 µm

0.308 - 2.1 µm

0.248 - 2.1 µm

0.308 - 2.1 µm

1064 nm, 7 ns

15 J

85 J

15 J

85 J

266 nm, 7 ns

MAX MEASURABLE ENERGYENERGY DETECTORSQE50LP-S-MB-QED

15 J

POWER DETECTORSQE50LP-S-MBa

Maximum measurable energy a
22 J

12.5 J

22 J

10 µJ

20 µJ

10 µJ

20 µJ

Max repetition frequency

200 Hz200 Hz200 Hz200 HzMaximum pulse width (typical) d

675 µs675 µs675 µs675 µsCalibration uncertainty

± 3%

± 3%

± 3%

± 3%

< 0.5 %

< 0.5 %

< 0.5 %

< 0.5 %

10 W

25 W

20 W

45 W

e

Repeatability
DAMAGE THRESHOLDS
Maximum average power

BEAM PROFILING

12.5 J

Noise equivalent energy c

Maximum energy density
0.6 J/cm2

16 J/cm2

0.6 J/cm2

16 J/cm2

1064 nm, 7 ns, 10 Hz

2

0.6 J/cm

2

8 J/cm2

0.6 J/cm

8 J/cm2

532 nm, 7 ns, 10 Hz

2

0.6 J/cm

6 J/cm2

0.6 J/cm

6 J/cm2

0.5 J/cm2

1 J/cm2

0.5 J/cm2

1 J/cm2

10 W/cm

600 W/cm

10 W/cm

600 W/cm2

266 nm, 7 ns, 10 Hz
Maximum average power density
PHYSICAL CHARACTERISTICS
Effective aperture

f

2

2

2

2

47 x 47 mm

50 x 50 mm

47 x 47 mm

MB

QED

MB

QED

Dimensions

75H x 75W x 15D mm

78H x 80W x 20D mm

75H x 75W x 44D mm

78H x 80W x 49D mm

Weight

209 g

209 g

338 g

338 g

DB15, USB or RS-232

DB15, USB or RS-232

DB15, USB or RS-232

DB15, USB or RS-232

STAND-D-233

STAND-D-233

STAND-D-233

STAND-D-233

ORDERING INFORMATION
Available output options
Compatible standProduct page

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com.com

CUSTOM / OEM PRODUCTS

a. Not exceeding maximum average power. Increasing pulse width increases the maximum measurable energy. The maximum measurable energy depends on the display or PC interface used.
If your laser is close to the maximum, contact us to check your specifications.
b. Calibration at 2.1 to 2.5 µm is available on special request.
c. Nominal value, actual value depends on electrical noise in the measurement system.
d. Also available on special order: ELP (extra-long pulse) version.
e. Excludes non-linearities.
f. At maximum power.

DISPLAYS & PC INTERFACES

50 x 50 mm

Absorber

TERAHERTZ DETECTORS1064 nm, 7 ns, single shot81

QE65-MB
65 x 65 mm, 10 µJ - 200 J

OUTPUT OPTIONSSMART INTERFACE
Containing all the calibration data
ALL-IN-ONE-METER
Connects directly to a PC
Three models available:
USB output (-INT)
RS-232 output (-IDR)
USB with external trigger (-INE)

COMPATIBLE DISPLAYS & PC INTERFACESKEY FEATURESMIRO ALTITUDE

MAESTRO

M-LINK

S-LINK

U-LINKACCESSORIESMODULAR CONCEPT
Increase the power capability of your detector:
2 different cooling modules
LARGE APERTURE
Effective aperture of 65 x 65 mm
QED ATTENUATOR AVAILABLE

Stand with delrin post
(200428, For -S model)Stand with delrin post(201284, For -H model)

QED-65 attenuator

Pelican carrying case

• Measure up to 5X higher energies• Available with optional calibration,
all wavelengths between 532 & 1064 nm,
or single wavelength
LOW NOISE LEVEL
10 µJ for the MB coating
TEST TARGET INCLUDED
With the MB models

82

gentec-eo.com/laser-energy-meter

DB15 to BNC adaptor

QE65-MB
Specifications

QE65LP-S-MB-QED QE65LP-H-MB

QE65LP-H-MB-QED QE65ELP-H-MB

25 J

200 J

25 J

200 J

50 J

MAX REPETITION FREQUENCY

100 Hz100 Hz100 Hz100 Hz

20 Hz

EFFECTIVE APERTURE

65 x 65 mm

62 x 62 mm

65 x 65 mm

62 x 62 mm

65 x 65 mm

Spectral range

0.19 - 20 µm

0.3 - 2.1 µm

0.19 - 20 µm

0.3 - 2.1 µm

0.19 - 20 µm

Calibrated spectral range b

0.248 - 2.1 µm

0.308 - 2.1 µm

0.248 - 2.1 µm

0.308 - 2.1 µm

0.248 - 2.1 µm

1064 nm, 150 µs

25 J

200 J

25 J

200 J

50 J

1064 nm, 7 ns

25 J

125 J

25 J

125 J

25 J

266 nm, 7 ns

20 J

35 J

20 J

35 J

20 J

MAX MEASURABLE ENERGY

POWER DETECTORSQE65LP-S-MBa

MEASUREMENT CAPABILITY

Noise equivalent energy

ENERGY DETECTORS

Maximum measurable energy

a

20 µJ

10 µJ

20 µJ

20 µJ

100 Hz100 Hz100 Hz100 Hz

20 Hz

Maximum pulse width (typical) d

0.7 ms

0.7 ms

0.7 ms

0.7 ms

5 ms

Calibration uncertainty

± 3%

± 3%

± 3%

± 3%

± 3%

< 0.5 %

< 0.5 %

< 0.5 %

< 0.5 %

< 0.5 %

12 W

30 W

40 W

90 W

40 W

e

Repeatability

BEAM PROFILING

10 µJ

Max repetition frequencyc

DAMAGE THRESHOLDS
Maximum average power
Maximum energy density
1.2 J/cm2

14 J/cm2

1.2 J/cm2

14 J/cm2

1.2 J/cm2

1064 nm, 7 ns, single shot2

0.6 J/cm

16 J/cm2

0.6 J/cm

16 J/cm

0.6 J/cm2

1064 nm, 7 ns, 10 Hz

0.6 J/cm2

8 J/cm2

0.6 J/cm2

8 J/cm2

0.6 J/cm2

532 nm, 7 ns, 10 Hz

0.6 J/cm2

6 J/cm2

0.6 J/cm2

6 J/cm2

0.6 J/cm2

266 nm, 7 ns, 10 Hz

0.5 J/cm

1 J/cm

0.5 J/cm

1 J/cm

0.5 J/cm2

10 W/cm2

600 W/cm2

10 W/cm2 h

600 W/cm2

10 W/cm2

Effective aperture

65 x 65 mm

62 x 62 mm

65 x 65 mm

62 x 62 mm

65 x 65 mm

Absorber

MB

QED

MB

QED

MB

Dimensions

92H x 92W x 20D mm

95H x 97W x 25D mm

92H x 92W x 99D mm

95H x 97W x 104D mm

92H x 92W x 99D mm

Weight

440 g

440 g

900 g

900 g

900 g

Available output options

DB15, USB or RS-232

DB15, USB or RS-232

DB15, USB or RS-232

DB15, USB or RS-232

DB15, USB or RS-232

Compatible stand

STAND-D-233

STAND-D-233

STAND-D-443

STAND-D-443

STAND-D-443

Maximum average power density f

2

2

2

2

2

2

TERAHERTZ DETECTORS1064 nm, 150 µs, 10 Hz

PHYSICAL CHARACTERISTICSDISPLAYS & PC INTERFACESORDERING INFORMATION

Product page

CUSTOM / OEM PRODUCTS

a. Not exceeding maximum average power. Increasing pulse width increases the maximum measurable energy. The maximum measurable energy depends on the display or PC interface used.
If your laser is close to the maximum, contact us to check your specifications.
b. Calibration at 2.1 to 2.5 µm is available on special request.
c. Nominal value, actual value depends on electrical noise in the measurement system.
d. Also available on special order: ELP (extra-long pulse) version.
e. Excludes non-linearities.
f. At maximum power.

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

83

QE95-MB
95 mm Ø, 15 µJ - 250 J

OUTPUT OPTIONSSMART INTERFACE
Containing all the calibration data
ALL-IN-ONE-METER
Connects directly to a PC
Three models available:
USB output (-INT)
RS-232 output (-IDR)
USB with external trigger (-INE)

COMPATIBLE DISPLAYS & PC INTERFACESKEY FEATURESMIRO ALTITUDE

MAESTRO

M-LINK

S-LINK

U-LINKACCESSORIESMODULAR CONCEPT
Increase the power capability of your detector:
2 different cooling modules
EXTRA LARGE APERTURE
Effective aperture of 95 mm Ø
QED ATTENUATOR AVAILABLE

Stand with delrin post
(200428, For -S model)Stand with delrin post(201284, For -H model)

QED-95 attenuator

Pelican carrying case

• Measure up to 5X higher energies• Available with optional calibration,
all wavelengths between 532 & 1064 nm,
or single wavelength
LOW NOISE LEVELTEST TARGET INCLUDED

84

gentec-eo.com/laser-energy-meter

DB15 to BNC adaptor

QE95-MB
Specifications

QE95LP-H-MB-QED QE95ELP-H-MB

250 J

35 J

250 J

70 J

MAX REPETITION FREQUENCY

40 Hz

40 Hz

40 Hz

40 Hz

10 Hz

EFFECTIVE APERTURE

95 mm Ø90 mm Ø95 mm Ø90 mm Ø95 mm ØMEASUREMENT CAPABILITY
Spectral range

0.19 - 20 µm

0.3 - 2.1 µm

0.19 - 20 µm

0.3 - 2.1 µm

0.19 - 20 µm

Calibrated spectral range b

0.248 - 2.1 µm

0.308 - 2.1 µm

0.248 - 2.1 µm

0.308 - 2.1 µm

0.248 - 2.1 µm

1064 nm, 150 µs

35 J

250 J

35 J

250 J

70 J

1064 nm, 7 ns

35 J

150 J

35 J

150 J

35 J

266 nm, 7 ns

30 J

50 J

30 J

50 J

30 J

Noise equivalent energy c

15 µJ

30 µJ

15 µJ

30 µJ

30 µJ

Max repetition frequency

40 Hz

40 Hz

40 Hz

40 Hz

10 Hz

ENERGY DETECTORSQE95LP-S-MB-QED QE95LP-H-MB

35 J

POWER DETECTORSQE95LP-S-MB
MAX MEASURABLE ENERGY a

Maximum measurable energy a

Maximum pulse width (typical)

1.5 ms

1.5 ms

1.5 ms

5 ms

± 3%

± 3%

± 3%

± 3%

± 3%

Repeatability< 0.5 %

< 0.5 %

< 0.5 %

< 0.5 %

< 0.5 %

20 W

45 W

40 W

90 W

40 W

1064 nm, 150 µs, 10 Hz

1.2 J/cm2

14 J/cm2

1.2 J/cm2

14 J/cm2

1.2 J/cm2

1064 nm, 7 ns, single shot

0.6 J/cm2

16 J/cm2

0.6 J/cm2

16 J/cm2

0.6 J/cm2

1064 nm, 7 ns, 10 Hz

2

0.6 J/cm

2

8 J/cm2

0.6 J/cm

2

8 J/cm

0.6 J/cm2

532 nm, 7 ns, 10 Hz

2

0.6 J/cm

2

6 J/cm2

0.6 J/cm

2

6 J/cm

0.6 J/cm2

266 nm, 7 ns, 10 Hz

2

0.5 J/cm

2

1 J/cm2

0.5 J/cm

2

1 J/cm

0.5 J/cm2

10 W/cm2

600 W/cm2

10 W/cm2 h

600 W/cm2

10 W/cm2

DAMAGE THRESHOLDS
Maximum average power

BEAM PROFILING

1.5 msCalibration uncertainty edMaximum energy density

PHYSICAL CHARACTERISTICS
Effective aperture

90 mm Ø95 mm Ø90 mm Ø95 mm Ø

MB

QED

MB

QED

MB

Dimensions

122H x 122W x 20D mm

125H x 127W x 25D mm

122H x 122W x 99D mm

125H x 127W x 104D mm

122H x 122W x 99D mm

Weight0.78 kg

0.78 kg

1.2 kg

1.2 kg

1.2 kgDB15, USB or RS-232

DB15, USB or RS-232

DB15, USB or RS-232

DB15, USB or RS-232

DB15, USB or RS-232

STAND-D-233

STAND-D-233

STAND-D-443

STAND-D-443

STAND-D-443

ORDERING INFORMATION
Available output options
Compatible stand

DISPLAYS & PC INTERFACES

95 mm ØAbsorber

TERAHERTZ DETECTORS

Maximum average power density f

Product page

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com.com

CUSTOM / OEM PRODUCTS

a. Not exceeding maximum average power. Increasing pulse width increases the maximum measurable energy. The maximum measurable energy depends on the display or PC interface used.
If your laser is close to the maximum, contact us to check your specifications.
b. Calibration at 2.1 to 2.5 µm is available on special request.
c. Nominal value, actual value depends on electrical noise in the measurement system.
d. Also available on special order: ELP (extra-long pulse) version.
e. Excludes non-linearities.
f. At maximum power.

85

QE145-MB

Extra-large pyroelectric energy detector

t

c
du

ew

o
pr

N

OUTPUT OPTIONSSMART INTERFACE
Containing all the calibration data
ALL-IN-ONE-METER
Connects directly to a PC
Three models available:
USB output (-INT)
RS-232 output (-IDR)
USB with external trigger (-INE)

COMPATIBLE DISPLAYS & PC INTERFACESKEY FEATURESMIRO ALTITUDE

MAESTRO

M-LINK

S-LINK

U-LINKACCESSORIESCUSTOM-BUILTContact us with your requirements for a version
tailored to your needs
MODULAR CONCEPT
Increase the power capability of your detector:
2 different cooling modules
EXTRA LARGE APERTURE
Effective aperture of 145 mm Ø
QED ATTENUATOR AVAILABLE• Measure up to 5X higher energies• Available with optional calibration,
all wavelengths between 532 & 1064 nm,
or single wavelength
LOW NOISE LEVELTEST TARGET INCLUDED

86

gentec-eo.com/laser-energy-meterStand with delrin post

DB15 to BNC adaptor

Pelican carrying case

QE145-MB
Specifications

Up to 385 J

MAX REPETITION FREQUENCY

Up to 200 Hz

Up to 200 Hz

EFFECTIVE APERTURE

145 mm Ø

135 mm Ø

MEASUREMENT CAPABILITY
Spectral range

0.19 - 20 µm

0.3 – 2.1 µm

Calibrated spectral range b

0.248 - 2.1 µm

0.308 - 2.1 µm

1064 nm, 150 µs

Up to 135 J

Up to 385 J

1064 nm, 7 ns

ENERGY DETECTORSQE145-QED, CUSTOM CAPABILITIESUp to 135 J

POWER DETECTORSQE145, CUSTOM CAPABILITIESMAX MEASURABLE ENERGY a

Maximum measurable energy a
Up to 220 J

As low as 100 µJ

As low as 200 µJ

Max repetition frequency

Up to 200 Hz

Up to 200 Hz

Maximum pulse width (typical)

Up to 5 ms

Up to 5 ms

Calibration uncertainty

±3%

±3%

< ± 0.5 %< ± 0.5 %Up to 75 W

Up to 190 W

1064 nm, 150 µs, 10 Hz

1.2 J/cm2

14 J/cm2

1064 nm, 7 ns, single shot2

0.6 J/cm

16 J/cm2

1064 nm, 7 ns, 10 Hz

0.6 J/cm2

8 J/cm2

266 nm, 7 ns, 10 Hz

0.5 J/cm

1 J/cm2e

Repeatability
DAMAGE THRESHOLDS
Maximum average power

BEAM PROFILING

Up to 70 J

Noise equivalent energy c

Maximum energy density

600 W/cm2

145 mm Ø

135 mm Ø

Absorber

MB or MT

QED

Dimensions

180H x 180W x 24D mm (thicker with heatsink)

180H x 180W x 27D mm (thicker with heatsink)

Weight

1.9 – 2.5 kg

2.3 – 2.8 kg

Cooling

Convection or heatsink

Convection or heatsink

DB15, USB or RS-232

DB15, USB or RS-232

STAND-D-443

STAND-D-443

PHYSICAL CHARACTERISTICS
Effective aperture

ORDERING INFORMATION
Available output options
Compatible stand2

DISPLAYS & PC INTERFACES

10 W/cmf

TERAHERTZ DETECTORS

Maximum average power density

2

Product pagea. Not exceeding maximum average power. Increasing pulse width increases the maximum measurable energy. The maximum measurable energy depends on the display or PC interface used.
If your laser is close to the maximum, contact us to check your specifications.
b. Calibration at 2.1 to 2.5 μm is available on special request.
c. Nominal value, actual value depends on electrical noise in the measurement system.
d. Excludes non-linearities.
e. At 12 W.

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com.com

CUSTOM / OEM PRODUCTS* These products are custom-built. Contact us with your requirements for a version tailored to your needs.

87

QE195-MB

The largest pyroelectric energy detector on the market

t

c
du

ew

o
pr

N

OUTPUT OPTIONSSMART INTERFACE
Containing all the calibration data
ALL-IN-ONE-METER
Connects directly to a PC
Three models available:
USB output (-INT)
RS-232 output (-IDR)
USB with external trigger (-INE)

COMPATIBLE DISPLAYS & PC INTERFACESKEY FEATURESMIRO ALTITUDE

MAESTRO

M-LINK

S-LINK

U-LINKACCESSORIESCUSTOM-BUILTContact us with your requirements for a version
tailored to your needs
MODULAR CONCEPT
Increase the power capability of your detector:
2 different cooling modules
EXTRA LARGE APERTURE
Effective aperture of 195 mm Ø
QED ATTENUATOR AVAILABLE• Measure up to 5X higher energies• Available with optional calibration,
all wavelengths between 532 & 1064 nm,
or single wavelength
LOW NOISE LEVELTEST TARGET INCLUDED

88

gentec-eo.com/laser-energy-meterStand with delrin post

DB15 to BNC adaptor

Pelican carrying case

QE195-MB
Specifications

Up to 700 J

MAX REPETITION FREQUENCY

Up to 200 Hz

Up to 200 Hz

EFFECTIVE APERTURE

195 mm Ø

185 mm Ø

MEASUREMENT CAPABILITY
Spectral range

0.19 - 20 µm

0.3 – 2.1 µm

Calibrated spectral range b

0.248 - 2.1 µm

0.308 - 2.1 µm

1064 nm, 150 µs

Up to 250 J

Up to 700 J

1064 nm, 7 ns

MAX MEASURABLE ENERGYENERGY DETECTORSQE195-QED, CUSTOM CAPABILITIESUp to 250 J

POWER DETECTORSQE195, CUSTOM CAPABILITIESaMaximum measurable energy a
Up to 400 J

As low as 100 µJ

As low as 200 µJ

Max repetition frequency

Up to 200 Hz

Up to 200 Hz

Maximum pulse width (typical) d

Up to 5 ms

Up to 5 ms

Calibration uncertainty

±3%

±3%

< ± 0.5 %< ± 0.5 %Up to 150 W

Up to 350 W

1064 nm, 150 µs, 10 Hz

1.2 J/cm2

14 J/cm2

1064 nm, 7 ns, single shot

0.6 J/cm2

16 J/cm2

1064 nm, 7 ns, 10 Hz

0.6 J/cm2

8 J/cm2

266 nm, 7 ns, 10 Hz

0.5 J/cm

1 J/cm2e

Repeatability
DAMAGE THRESHOLDS
Maximum average power

BEAM PROFILING

Up to 125 J

Noise equivalent energy c

Maximum energy density

600 W/cm2

195 mm Ø

185 mm Ø

Absorber

MB or MT

QED

Dimensions

229H x 229W x 24D mm (thicker with heatsink)

229H x 229W x 27D mm (thicker with heatsink)

Weight3 - 5 kg

3 - 6 kg

Cooling

Convection or heatsink

Convection or heatsink

DB15, USB or RS-232

DB15, USB or RS-232

STAND-D-443

STAND-D-443

PHYSICAL CHARACTERISTICS
Effective aperture

ORDERING INFORMATION
Available output options
Compatible stand2

DISPLAYS & PC INTERFACES

10 W/cmf

TERAHERTZ DETECTORS

Maximum average power density

2

Product pagea. Not exceeding maximum average power. Increasing pulse width increases the maximum measurable energy. The maximum measurable energy depends on the display or PC interface used.
If your laser is close to the maximum, contact us to check your specifications.
b. Calibration at 2.1 to 2.5 μm is available on special request.
c. Nominal value, actual value depends on electrical noise in the measurement system.
d. Excludes non-linearities.
e. At 12 W.

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com.com

CUSTOM / OEM PRODUCTS* These products are custom-built. Contact us with your requirements for a version tailored to your needs.

89

QE12-MT

12 x 12 mm, 0.7 µJ - 3.9 J, tuned for high repetition rates

OUTPUT OPTIONSSMART DB15 CONNECTOR
Contains all the calibration data
ALL-IN-ONE-METER
Connects directly to a PC
Three models available: 				USB output (-INT)
RS-232 output (-IDR)
USB with external trigger (-INE)

COMPATIBLE DISPLAYS & PC INTERFACESKEY FEATURESMIRO ALTITUDE

MAESTRO

M-LINK

S-LINK

U-LINKACCESSORIESMODULAR CONCEPT
Increase the power capability of your detector:
2 different cooling modules
LOW NOISE LEVELNEW MODELS FOR HIGH REPETITION RATESThe QE12HR models are tuned for short pulses withhigh repetition rates (up to 10 kHz)Stand with delrin post

Pelican carrying case90

gentec-eo.com/laser-energy-meter

DB15 to BNC adaptor

QED-12 attenuator

QE12-MT
Specifications

MAX REPETITION FREQUENCYb, c

QE12SP-S-MT-D0

QE12SP-H-MT-D0

QE12HR-H-MT-D0

0.70 J

0.70 J

0.70 J

6 kHz

10 kHz

12 x 12 mm

12 x 12 mm

12 x 12 mm

MEASUREMENT CAPABILITY
Spectral range
Calibrated spectral range d

0.19 - 20 µm

0.19 - 20 µm

0.19 - 20 µm

0.248 - 2.1 µm

0.248 - 2.1 µm

0.248 - 2.1 µm

1064 nm, 7 ns

0.70 J

0.70 J

0.70 J

266 nm, 7 ns

0.10 J

0.10 J

0.10 J

Noise equivalent energy e

0.8 µJ

0.8 µJ

1 µJ

Max repetition frequency b, c

6 kHz

6 kHz

10 kHzMaximum pulse width (typical)

10 µs

10 µs

4 µs

Calibration uncertainty f

± 3%

± 3%

± 3%

Repeatability< 0.5 %

< 0.5 %

< 0.5 %

3W

5W

5W

1064 nm, 7 ns, single shot

0.50 J/cm2

0.50 J/cm2

0.50 J/cm2

1064 nm, 7 ns, 10 Hz

0.50 J/cm2

0.50 J/cm2

0.50 J/cm2

532 nm, 7 ns, 10 Hz

0.07 J/cm2

0.07 J/cm2

0.07 J/cm2

266 nm, 7 ns, 10 Hz

0.07 J/cm

0.07 J/cm

0.07 J/cm2

10 W/cm2

10 W/cm

10 W/cm2

Effective aperture
Absorber

12 x 12 mm

12 x 12 mm

12 x 12 mm

MT

MT

MT

Dimensions

36H x 36W x 14D mm

36H x 36W x 33D mm

36H x 36W x 33D mm

Weight

87 g

117 g

117 g

DB15, USB or RS-232

DB15, USB or RS-232

DB15, USB or RS-232

STAND-D-233

STAND-D-233

STAND-D-233

ENERGY DETECTORS

6 kHz

APERTURE

POWER DETECTORSMAX MEASURABLE ENERGYaMaximum measurable energy a

Maximum average power
Maximum energy density

Maximum average power density

2

g

2

2

BEAM PROFILINGDAMAGE THRESHOLDSTERAHERTZ DETECTORSPHYSICAL CHARACTERISTICS

ORDERING INFORMATION
Available output options
Compatible standProduct page

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com.com

CUSTOM / OEM PRODUCTS

PRODUCT GUIDE 2024

DISPLAYS & PC INTERFACES

a. Not exceeding maximum average power. Increasing pulse width increases the maximum measurable energy. The maximum measurable energy depends on the display or PC interface used.
If your laser is close to the maximum, contact us to check your specifications.
b. With the IDR version, measured values are sampled when the repetition rate is > 200 Hz.
c. Maximum 5.2 kHz with INT version.
d. Calibration at 2.1 to 2.5 µm is available on special request.
e. Nominal value, actual value depends on electrical noise in the measurement system.
f. Excludes non-linearities.
g. At maximum power.

91

QE25-MT

25 x 25 mm, 2 µJ - 23 J, tuned for high repetition rates

OUTPUT OPTIONSSMART DB15 CONNECTOR
Contains all the calibration data
ALL-IN-ONE-METER
Connects directly to a PC
Three models available: 				USB output (-INT)
RS-232 output (-IDR)
USB with external trigger (-INE)

COMPATIBLE DISPLAYS & PC INTERFACESKEY FEATURESMIRO ALTITUDE

MAESTRO

M-LINK

S-LINK

U-LINKACCESSORIESMODULAR CONCEPT
Increase the power capability of your detector:
2 different cooling modules
LOW NOISE LEVELNEW MODELS FOR HIGH REPETITION RATESThe QE25HR models are tuned for short pulses withhigh repetition rates (up to 10 kHz)Stand with delrin post

Pelican carrying case92

gentec-eo.com/laser-energy-meter

DB15 to BNC adaptor

QED-25 attenuator

QE25-MT
Specifications

QE25SP-H-MT-D0

QE25HR-H-MT-D0

3.0 J

3.0 J

3.0 J

MAX REPETITION FREQUENCY b, c

6 kHz

6 kHz

10 kHz

EFFECTIVE APERTURE

25 x 25 mm

25 x 25 mm

25 x 25 mm

Spectral range

0.19 - 20 µm

0.19 - 20 µm

0.19 - 20 µm

Calibrated spectral range d

0.248 - 2.1 µm

0.248 - 2.1 µm

0.248 - 2.1 µm

1064 nm, 7 ns

3.0 J

3.0 J

3.0 J

266 nm, 7 ns

0.44 J

0.44 J

0.44 J

Noise equivalent energy e

2 µJ

2 µJ

3 µJ

Max repetition frequency b, c

6 kHz

6 kHz

10 kHzMaximum pulse width (typical)

10 µs

10 µs

4 µs

Calibration uncertainty

± 3%

± 3%

± 3%

< 0.5 %

< 0.5 %

< 0.5 %

5W

10 W

10W

1064 nm, 7 ns, single shot

0.50 J/cm2

0.50 J/cm2

0.50 J/cm2

1064 nm, 7 ns, 10 Hz

0.50 J/cm2

0.50 J/cm2

0.50 J/cm2

532 nm, 7 ns, 10 Hz

0.07 J/cm2

0.07 J/cm2

0.07 J/cm2

266 nm, 7 ns, 10 Hz

0.07 J/cm

0.07 J/cm

0.07 J/cm2

10 W/cm

10 W/cm

10 W/cm2

MAX MEASURABLE ENERGY

POWER DETECTORSQE25SP-S-MT-D0a

MEASUREMENT CAPABILITY
ENERGY DETECTORS

Maximum measurable energy af

DAMAGE THRESHOLDS
Maximum average power

BEAM PROFILING

Repeatability

Maximum energy density

PHYSICAL CHARACTERISTICS
Effective aperture

g

2

2

2

25 x 25 mm

25 x 25 mm

25 x 25 mm

Absorber

MT

MT

MT

Dimensions

50H x 50W x 14D mm

50H x 50W x 53D mm

50H x 50W x 53D mm

Weight

193 g

193 g

193 g

Available output options

DB15, USB or RS-232

DB15, USB or RS-232

DB15, USB or RS-232

Compatible stand

STAND-D-233

STAND-D-233

STAND-D-233

TERAHERTZ DETECTORS

Maximum average power density

2

ORDERING INFORMATIONPRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com.com

CUSTOM / OEM PRODUCTS

a. Not exceeding maximum average power. Increasing pulse width increases the maximum measurable energy. The maximum measurable energy depends on the display or PC interface used.
If your laser is close to the maximum, contact us to check your specifications.
b. With the IDR version, measured values are sampled when the repetition rate is > 200 Hz.
c. Maximum 5.2 kHz with INT version.
d. Calibration at 2.1 to 2.5 µm is available on special request.
e. Nominal value, actual value depends on electrical noise in the measurement system.
f. Excludes non-linearities.
g. At maximum power.

DISPLAYS & PC INTERFACES

Product page93

QE50-MT
50 x 50 mm, 10 µJ - 85 J

OUTPUT OPTIONSSMART DB15 CONNECTOR
Contains all the calibration data
ALL-IN-ONE-METER
Connects directly to a PC
Three models available: 				USB output (-INT)
RS-232 output (-IDR)
USB with external trigger (-INE)

COMPATIBLE DISPLAYS & PC INTERFACESMIRO ALTITUDE

MAESTRO

M-LINK

S-LINK

U-LINKKEY FEATURESMODULAR CONCEPT
Increase the power capability of your detector:
2 different cooling modules
LOW NOISE LEVELACCESSORIESQED ATTENUATOR AVAILABLE
Measure up to 5X higher energies
Available with optional calibration, allwavelengths between 532 & 1064 nm, or single
wavelength
HIGH REPETITION RATE
Measure each pulse at up to 4000 Hz

Stand with delrin post

Pelican carrying case94

gentec-eo.com/laser-energy-meter

DB15 to BNC adaptor

QED-50 attenuator

QE50-MT
Specifications

13 J

MAX REPETITION FREQUENCY b, c

4000 Hz

4000 Hz

EFFECTIVE APERTURE

50 x 50 mm

50 x 50 mm

MEASUREMENT CAPABILITY
Spectral range

0.19 - 20 µm

0.19 - 20 µm

Calibrated spectral range d

0.248 - 2.1 µm

0.248 - 2.1 µm

1064 nm, 7 ns

13 J

13 J

266 nm, 7 ns

MAX MEASURABLE ENERGYENERGY DETECTORSQE50SP-H-MT-D0

13 J

POWER DETECTORSQE50SP-S-MT-D0aMaximum measurable energy a
1.8 J

10 µJ

10 µJ

Max repetition frequency b, c

4000 Hz

4000 Hz

Maximum pulse width (typical)

10 µs

10 µs

Calibration uncertainty

± 3%

± 3%

< 0.5 %

< 0.5 %

10W

20W

f

Repeatability
DAMAGE THRESHOLDS
Maximum average power

BEAM PROFILING

1.8 J

Noise equivalent energy e

Maximum energy density
0.50 J/cm2

1064 nm, 7 ns, 10 Hz

20.50 J/cm

0.50 J/cm2

532 nm, 7 ns, 10 Hz

0.07 J/cm2

0.07 J/cm2

266 nm, 7 ns, 10 Hz

0.07 J/cm

0.07 J/cm2

10 W/cm

10 W/cm2

50 x 50 mm

50 x 50 mm

AbsorberMB

MB

Dimensions

75H x 75W x 15D mm

75H x 75W x 44D mm

Weight

209 g

338 g

DB15, USB or RS-232

DB15, USB or RS-232

STAND-D-233

STAND-D-233

Maximum average power density
PHYSICAL CHARACTERISTICS
Effective aperture

ORDERING INFORMATION
Available output options
Compatible stand

2

g

2

Product page

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com.com

CUSTOM / OEM PRODUCTS

a. Not exceeding maximum average power. Increasing pulse width increases the maximum measurable energy. The maximum measurable energy depends on the display or PC interface used.
If your laser is close to the maximum, contact us to check your specifications.
b. With the IDR version, measured values are sampled when the repetition rate is > 200 Hz.
c. Maximum 5.2 kHz with INT version.
d. Calibration at 2.1 to 2.5 µm is available on special request.
e. Nominal value, actual value depends on electrical noise in the measurement system.
f. Excludes non-linearities.
g. At maximum power.

DISPLAYS & PC INTERFACES

0.50 J/cm2

TERAHERTZ DETECTORS1064 nm, 7 ns, single shot

95

IS50

Integrating sphere detector for laser energy measurement up to 1 kW

t
uc

ew

od
r
p

N

OUTPUT OPTIONSSMART DB15 CONNECTOR
Contains all the calibration data
ANALOG OUTPUT
When used with APM (D) analog power supply
ALL-IN-ONE-METER
Connects directly to a PC
Three models available: 				USB output (-INT)
RS-232 output (-IDR)
USB with external trigger (-INE)

COMPATIBLE DISPLAYS & PC INTERFACES

MAESTROU-LINK

M-LINKKEY FEATURES
S-LINKCUSTOM-BUILTContact us with your requirements for a version
tailored to your needs

ACCESSORIESWIDE ENERGY RANGE
Very low noise level = wide energy range withjust one device
HIGH AVERAGE POWER
The water-cooled joulemeters can handle highaverage powers, up to 1000 W.

Stand with delrin post

DB15 to BNC adaptor

RESISTANT COATING
Our proprietary coating is designed to be strong.
Its damage thresholds are orders of magnitude
higher than any other “white” coatings on themarket.
PRECISE CALIBRATIONThe IS detectors have a NIST-traceablecalibration for the entire calibrated spectral
range.

96

gentec-eo.com/laser-energy-meter

Fiber adaptors & connectors(FC, ST or SMA)

APM (D)
analog power supply

Pelican carrying caseIS50

Specifications

POWER DETECTORSIS50A-1KW-RMTMAX MEASURABLE ENERGY a

1 J, can be higher for low repetition rates

MAX REPETITION FREQUENCY

1000 Hz

EFFECTIVE APERTURE

50 mm ØCOOLING METHOD

Water

MEASUREMENT CAPABILITY
532 nm, 1064 nm, and 1070 nm, other wavelengths available on request

Maximum measurable energy
1064 nm, 7 ns

1 J, can be higher for low repetition rates

Noise equivalent energy b< 1 mJ

Max repetition frequency

1000 Hz

Maximum pulse width (typical)

10 µs

Maximum divergence

10° (half-angle)

Maximum incidence angle± 25° for beam diameter < Ø 12mm
± 5° for beam diameter > Ø 12mm

Calibration uncertainty c±4%Back reflections

12 %, concentrated in a cone with 15 degrees half-angle

Repeatability< ± 0.5 %BEAM PROFILING

0.19 – 2.5 µm

Calibrated spectral range

ENERGY DETECTORS

Spectral range

DAMAGE THRESHOLDS
1000 W

Maximum energy density
532 nm and 1064 nm, 7 ns

400 mJ/cm2

Maximum average power density d

400 mJ/cm2

TERAHERTZ DETECTORS

Maximum average power

PHYSICAL CHARACTERISTICS
50 mm ØMounting thread

SM2

Sphere inner diameter

100 mm Ø

SensorPyroelectric

Dimensions

127H x 140W x 115D mm

Weight

4 kg

DISPLAYS & PC INTERFACES

Effective aperture

ORDERING INFORMATION
Available output options

DB15, USB or RS-232

Compatible stand

STAND-D-443

Product page

CUSTOM / OEM PRODUCTS* These products are custom-built. Contact us with your requirements for a version tailored to your needs.
a. Not exceeding maximum average power. Increasing pulse width increases the maximum measurable energy.
b. Nominal value, actual value depends on electrical noise in the measurement system.
c. Excludes non-linearities.
d. At 1064 nm.

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

97

PRONTO-500-IPL

Portable laser probe for IPL sources, 2 - 350 J per pulse

USER INTERFACE (SSE MODE)

Wait…

Trigger detected

Measurement complete!

The device waits
for a laser beam

Automatically starts when
exposed to a laser beam

The value is displayed
until the next measurement

Adjust the wavelength and calibration
Wavelength

Calibration

Warns you when
the device is too hot** Device may get hot, it is not recommended for handheld use when making a measurement

KEY FEATURESACCESSORIESHIGH ENERGY PER PULSE
Accurate readings up to 350 J/pulse!
EASY TO USE
The touchscreen color LCD allows for a friendly
user interface. You can make a measurementwith just the touch of a button!
DATA LOGGING
Save your data to the internal memory andthen transfer them to your PC over the USB
connection.
LARGE APERTURE
55 mm Ø aperture to accommodate large
beams
RUGGED• All-metal body• High damage thresholds
PROTECTIVE WINDOW• For measurements with gel-coupled IPL heads.
• Protects the absorber, easy to clean

SERIAL COMMANDS
Serial commands are available to let you takefull control of your PRONTO from your PC.
98

gentec-eo.com/laser-energy-meterStand with steel post

Pelican carrying caseSet the brightness
and orientation

PRONTO-500-IPL
Specifications

POWER DETECTORSPRONTO-500-IPL
MAX PULSE ENERGY

350 J

EFFECTIVE APERTURE

55 mm ØAPERTURE TYPE

Full aperture with protective window(SINGLE SHOT)

MEASUREMENT CAPABILITY
0.19 - 2.5 µm

Calibrated spectral range

1064 nm

Energy range

2 - 350 J

Noise equivalent energy

500 mJ

Minimum repetition period

15 s (= time between measurements)

Maximum pulse width

433 ms

Accuracy± 5%

ENERGY DETECTORS

Spectral range

DAMAGE THRESHOLDS
45 kW/cm2 (1064 nm, 10 W, CW)

Pulsed laser damage threshold

175 J/cm2 (10 ms pulses)

BEAM PROFILING

Maximum average power density

Maximum allowable absorber temperature 65 °C
GENERAL SPECIFICATIONSTouchscreen color LCD

Display size

28.0 x 35.0 mm (128 x 160 pixels)

Data storage

50 000 pts

Battery type

Rechargeable Li-ion

Battery life

17 hours or 4 200 measurements (with brightness set at 25 %)Battery recharge via

USB port

TERAHERTZ DETECTORS

Display type

PHYSICAL CHARACTERISTICS
Effective aperture

55 mm ØDimensions (sensor head)

88W x 88L x 32D mm (194L with handle)

Dimensions (monitor)

41W x 136L x 16D mm

Weight930 g

Compatible stand

DISPLAYS & PC INTERFACESORDERING INFORMATIONSTAND-S-443

Product page

CUSTOM / OEM PRODUCTS

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

99

CALORIMETERS

Photo courtesy of Lawrence Livermore National Laboratory

Measuring the highest energy laser beams

A Gentec-EO calorimeter is the only reliable solution available for thelargest and highest energy laser beams. Through cooperation withseveral leading research facilities around the world, Gentec-EO has
become the expert in manufacturing, calibrating and servicing
calorimeters for use in high energy inertial confinement fusion
calorimetric measurement.

100

gentec-eo.com/laser-energy-meter

POWER DETECTORSSTATE-OF -THE-ART
We work with a wide range of materials from surface coatings to the most robust
volume absorbers to provide the best solution for your specific application.

OUTSTANDING SIGNAL-TO-NOISE RATIOHIGH SENSITIVITYVACUUM COMPATIBILITY

With over 50 years of experience in thermal-based energy measurement, Gentec-EO
is the ideal choice for all your high energy measurement needs.

Using NIST traceable sources and proven calibration techniques, your Gentec-EO
calorimeter is always the most accurate large aperture measurement device on themarket.

TERAHERTZ DETECTORSWith calibration uncertainties of ± 3 %, and repeatabilities better than ± 2 % evenfor very large beams, Gentec-EO offers the very best solution for extreme energy
measurements.

DISPLAYS & PC INTERFACESCUSTOMIZEDWe have designed calorimeters for 16 kJ beams with apertures as large as 420 x
427 mm, and able to withstand pulse energy densities of more than 15 J/cm2.
We have also provided smaller, highly-sensitive calorimeters for beam energies as
low as 50 mJ for the most delicate applications.

T 418.651.8003 | info@gentec-eo.com

CUSTOM / OEM PRODUCTS

Our calorimeters range from 190 nm to 20 microns. Moreover, we are happy
to push these limits even further. We work with a wide range of materials fromsurface coatings to the most robust volume absorbers to provide the best solution
for your specific application.

PRODUCT GUIDE 2024

BEAM PROFILING

ACCURATE

ENERGY DETECTORSATTENTION TO DETAILS AND WORKMANSHIP

.com

101

CALORIMETERS
Applications

LASER FUSION EXPERIMENTS
Inertial confinement fusion (ICF) is a process where nuclear fusion reactions are initiated by heating and compressing a fuel target,
typically in the form of a pellet that most often contains a mixture of deuterium and tritium. To compress and heat the fuel, energy isdelivered to the outer layer of the target using high-energy beams of laser light. ICF is said to reproduce the energy generation process
taking place in the core of the sun.
Several laser fusion projects are underway around the world right now, their main goal is to produce a clean, reliable and nearly
unlimited source of energy. All these laser fusion experiments use very high energy lasers of sereval kJ per pulse for which a Gentec-EO
calorimeter is the ONLY reliable measuring device available on the market. Over the years, we have been presented with increasingly
large and energetic laser pulses to be measured and we have kept pace with the world’s most demanding lasers.

LASER FUSION MECHANISM

Schematic of the stages of inertial confinement fusion using lasers. The blue arrows represent radiation;orange is blowoff; purple is inwardly transported thermal energy.

2. 

3.

4.

Laser beams or laserproduced X-rays rapidly
heat the surface of thefusion target, forming
a surrounding plasma
envelope.

Fuel is compressed by therocket-like blowoff of the
hot surface material.

During the final part ofthe capsule implosion,
the fuel core reaches
20 times the density oflead and ignites at
100 000 000 ˚C.

Thermonuclear burnspreads rapidly throughthe compressed fuel,yielding many times the
input energy.

EXTREME PEAK POWER APPLICATIONS

Ultrashort pulsed lasers are developping at a very fast pace.
Some lasers now feature peak powers in the petawatts (1015 W).
Furthermore, the beam sizes can be fairly small, which results inpeak power densities too high for a standard detector. Typically,
pulse values for these lasers are in the range:
Beam sizes: up to 160 mm Ø
Energy range: 1 J to 100 J
Pulse widths: femtosecond & picosecond
Wavelengths: UV to NIR
For these, a Gentec-EO calorimeter is the only reliable solution.
Furthermore, it can sometimes be used in power mode.

102

gentec-eo.com/laser-energy-meter

Image courtesy of the University of Colorado and NSF

1.

CALORIMETERS
Technical aspects

EXAMPLES OF CUSTOM CALORIMETERSPULSE WIDTHMINIMUM ENERGY

MAXIMUM ENERGY*

427 mm x 420 mm

355 nm to 1064 nm

fs to ms

~100 J~5000 J

230 mm x 230 mm

355 nm to 1064 nm

fs to ms

~30 J

~1500 J

160 mm x 160 mm

355 nm to 1064 nm

fs to ms

~10 J

~750 J

120 mm x 120 mm

355 nm to 1064 nm

fs to ms

~5 J

~400 J

110 mm x 110 mm

355 nm to 1064 nm

fs to ms

~5 J

~350 J

310 mm Ø

355 nm to 1064 nm

fs to ms

~40 J

~2000 J

150 mm Ø

355 nm to 1064 nm

fs to ms

~5 J

~500 J

POWER DETECTORSSPECTRAL RANGERECTANGULAR APERTURES

ENERGY DETECTORSCIRCULAR APERTURES* For nanosecond pulses at 1064 nm. Maximum measurable energy depends on pulse width and wavelength. It generally decreases for shorter wavelengths and/or shorter pulse widths.

BEAM PROFILINGMONITORINGACQUISITIONREMOTE

INTERFACE

SOFTWARE

SYSTEM DIAGNOSTICS

Single-channel power & energy
PC interface (USB or RS-232)

Can handle several calorimeters saves data
to the PC graphic display

Validation of the calibration
Verification of the signal response

The P-LINK is the perfect PC interface to
be integrated into your system and used
remotely. You have the choice between
USB or RS-232 connection. The P-LINK
comes with a complete acquisition
software (PC-CALO).

The PC-Calo is a user-friendly PC
interface that reads and controls several
channels simultaneously. It reads thevoltage outputs of the PC interface,saves the data in a spreadsheet, displays
the data graphically and analyzes themeasured energy. The parameters areentered seperately and the data can betreated individually or simultaneously.

Do the on-site monitoring of yourcalorimeter using our special diagnostic
tool. The verification is done remotely so
you can control it from another location.
The diagnostic includes the verificationof the calorimeter’s calibration and of thesignal response and data acquisition.

TERAHERTZ DETECTORS

PC

.com

CUSTOM / OEM PRODUCTS

T 418.651.8003 | info@gentec-eo.com

DISPLAYS & PC INTERFACESPRODUCT GUIDE 2024

103

ABSORPTION CURVES
QE-MT & QE-MB
100%
90%
MB

80%

Absorptance

70%
60%
MT

50%
40%
30%
20%
10%
0%

0.1

1

10

Wavelength (µm)

0.25

10

0.20

8

0.15

6

0.10

4

0.05

2

0
200

400

600

800

1000

Wavelength (nm)104

gentec-eo.com/laser-energy-meter

1200

1400

1600

0

PE5B-GE Maximum Energy (nJ)

PE10B-SI Maximum Energy (µJ)

PE maximum energy

ABSORPTION CURVES
QED attenuators
POWER DETECTORS

100%
90%
80%
70%

ENERGY DETECTORSTransmittance

60%
50%
QED

40%
30%
20%

BEAM PROFILING

10%
0%

0.1

1

10

Wavelength (µm)

TERAHERTZ DETECTORS

QE-B & UM-B
100%
90%

BL

80%

AbsorptanceDISPLAYS & PC INTERFACES

70%
60%
50%

MT

40%
30%

CUSTOM / OEM PRODUCTS

20%
10%
0%

0.1

1

10

Wavelength (µm)

PRODUCT GUIDE 2024

T 418.651.8003 | info@gentec-eo.com

.com

105

BEAM PROFILING
Overview of the different models

BEAM PROFILING CAMERASProfiling a laser beam is a convenient complement to the measurementof its power or energy because it provides very useful additional
information, like spatial energy or intensity distribution, beam widths,
centroid, ellipticity and orientation, that may help you determine if yourlaser-based systems are operating optimally.
The Beamage is the most cost-effective USB3.0 Beam Profiling Camera on
the market. It is available for UV to IR wavelengths and in 2 sizes. It comeswith an intuitive and complete software that features an array of usefultools and functions. Its calculations are ISO compliant.

MAIN SPECIFICATIONSBEAMAGE-4M

BEAMAGE-4M-IR

BEAMAGE-4M-FOCUS

Camera only

350 - 1150 nm

1495 - 1595 nm

350 - 1150 nm

With UG11-UV filter

250 - 370 nm

---

---

With B3-IR-Filter

1250 - 1350 nm

---

---

Pixel count

4.2 MPixels

4.2 MPixels

4.2 MPixels

HxV

2048 x 2048

2048 x 2048

2048 x 2048

Sensor size

11.3 x 11.3 mm

11.3 x 11.3 mm

20.5 x 20.5 mm

Wavelength range

BEAM QUALITY MEASUREMENT
The performance of a laser in practical applications is critical in the designof optical systems and focusing applications, and it can be quantified bymeasuring M2, the laser beam quality factor, which indicates how close alaser is to being an ideal Gaussian beam.
The Beamage-M2 acquires a sequence of beam profile measurements toautomatically perform beam quality measurements within a few seconds.
It is equipped with the largest optics on the market for easy alignment
and fast measurements that you can trust. Its software is both intuitive
and ISO compliant.

106

gentec-eo.com/laser-beam-diagnostics

ACCESSORIESSpecifications

WAVELENGTH
BEAM SIZELASER POWER

YOU CAN MANAGE THEM WITH THE ACCESSORIES PRESENTED BELOW

POWER DETECTORS!

IF YOUR LASERSPECIFICATIONS EXCEEDTHE LIMITS IN TERMS OF

MANAGE THE WAVELENGTH
ENERGY DETECTORS

Since CMOS sensors are not sensitive to every frequency of the electromagnetic
spectrum, we offer several wavelength management solutions to enhance thecapabilities of the Beamage beam profiling cameras.

UV

VisibleIR

350 nm1150 nm

X-Rays(1495 nm - 1595 nm)
Beamage(no accessories needed)

Beamage+

BEAM PROFILING

Beamage+

IR Adaptor

UV ConverterorBeamage-IR

MANAGE THE BEAM SIZE

BeamageDiffuser

Gentec-EO detector
or beam dump

Camera Lens

Beamage

Sensor Plane

Camera Lens

Image ofthe Beam

Laser

DISPLAYS & PC INTERFACES

Sensor
Plane

Laser

Image ofthe Beam

MANAGE THE LASER POWER

CUSTOM / OEM PRODUCTS

CMOS sensors have low saturation levels as well as low damage thresholds. It is thus
very important that you control your laser power to get the best measurement possible
and avoid damaging the Beamage camera.
• For laser power under 1 W, you can attenuate the beam with ND filters• For laser power up to 1000 W, you can sample a small fraction of the beamwith a BA optical sampler

PRODUCT GUIDE 2024

T 418.651.8003 | info@gentec-eo.com

TERAHERTZ DETECTORS

A simple solution is offered to those who need to profile beams that are larger than the
CMOS sensor (> 11.3 mm x 11.3 mm). This solution is a beam reducing optical component
called camera lens. It works either by indirectly imaging the transmission of the beam after
it has passed through a diffusing element or by directly imaging the beam that is incident
on a Gentec-EO detector or beam dump.

.com

107

BEAMAGE

CMOS beam profiling cameras

INTUITIVE SOFTWARE INTERFACE
Easy to navigate interface, with many displays and control features:
2D, 3D and XY Displays
Background Subtraction Function
Unique “Animate” Function
Gaussian Fit
Semi-Log Graph

KEY FEATURESACCESSORIES

USB 3.0 FOR THE FASTEST TRANSFER RATESUp to 10X faster than regular USB 2.0
connections
HIGH RESOLUTION
4.2 Mpixels resolution gives accurate profile
measurements of very small beams
LARGE APERTURES

Stand with delrin post

BA series
optical attenuators

Stackable ND filters(0.5, 1.0, 2.0, 3.0, 4.0 & 5.0)

UV and IR filters

Pelican carrying case

Fiber adaptors & connectors(FC, ST and SMA)UV converters &IR adaptors

USB-A to USB-C adaptor• 11.3 x 11.3 mm for the Beamage-4M• 20.5 x 20.5 mm for the Beamage-4M-FOCUS
AVAILABLE WITH IR COATING
Beamage 4M-IR cameras have a special phosphor
coating for IR wavelengths (1495-1595 nm)
ISO COMPLIANT
D4Ơdefinition of diameter, centroid, ellipticity
and orientation are ISO 11146:2004 and 11146:2005
compliant
EXTERNAL TRIGGER
To synchronize the camera with a pulsed laser

108

gentec-eo.com/laser-beam-diagnostics

BEAMAGE
Specifications

BEAMAGE-4M-IR

BEAMAGE-4M-FOCUS

CMOS

CMOS

(with phosphor coating)(with fiber optic taper)

SENSOR TECHNOLOGYCMOS

EFFECTIVE APERTURE

11.3 x11.3 mm

11.3 x 11.3 mm

20.5 X 20.5 mm a

Camera only

350 - 1150 nm

1495 - 1595 nm

350 - 1150 nm

With UG11-UV filter

250 - 370 nm

---

---

With B3-IR-filter

POWER DETECTORSBEAMAGE-4M

MEASUREMENT CAPABILITY
Wavelength range

---

---

4.2 MPixels

4.2 MPixels

4.2 MPixels

HxV

2048 x 2048

2048 x 2048

2048 x 2048

Minimum measurable beam

55 µm

70 µm

120 µmRMS noise

1000:1 (60 dB)

1000:1 (60 dB)

1000:1 (60 dB)

DAMAGE THRESHOLDS
Maximum average power

1 W with ND filter

1 W with ND filter

1 W with ND filter

10 W/cm2
0.1 J/cm2

10 W/cm2
0.1 J/cm2

10 W/cm2
0.1 J/cm2

SOFTWARE
Displays

2D, 3D, XY and Beam Tracking
2D: Print Screen, Reset View, Show/Hide Beam Diameter
3D: Print Screen, Reset View, Top View
XY: Save Data, Zoom, Gaussian Fit, Semi-Log, Show/Hide Cursor, Show/Hide FWHM, Show/Hide 1/e2
Beam Tracking: Save Data, Print Screen, Reset View, Zoom

Display Features

Buffer Controls

Open File, Save Current Data, Save All Data, Previous/Next Image, Clear Buffer, Animate
Full Report in Print Ready Format (2D, 3D, XY, Measures, Parameters)
Print Screen in BMP format (2D and 3D)

Printing and Reports

11.3 x 11.3 mm

11.3 x 11.3 mm

Sensor area

1.28 cm

1.28 cm

1.28 cm2

Effective aperture

Same as sensor

Same as sensor

20.5 x 20.5 mm

Dimensions (not including filter)

61H x 81.1W x 19.7D mm

61H x 81.1W x 19.7D mm

61H x 81.1W x 46.5D mm

Weight (head only)

138 g

138 g

235 g

STAND-D-233

STAND-D-233

STAND-D-233

2

2

DISPLAYS & PC INTERFACES

11.3 x 11.3 mm

ORDERING INFORMATION
Compatible stand

TERAHERTZ DETECTORS

D4σ (ISO compliant),
1/e2 along crosshairs (13.5 %)FWHM along crosshairs (50 %)Custom ( %)Beam Diameter Definitions

PHYSICAL CHARACTERISTICS
Sensor size

BEAM PROFILING

Maximum density (1064 nm)

ENERGY DETECTORS

1250 - 1350 nm

Pixel count

Product page

CUSTOM / OEM PRODUCTS

a. With a typical pixel multiplication factor (PMF) of 1.8.

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

109

BEAMAGE
Software features

Capture Controls:
Start/Stop,
Subtract Background

Buffer Controls:
View Data Offline

Data:
Computation

File Controls:
Save and Open Data

Startup Configuration:
Save and Load Settings

1

2

Display Controls:
Print, Refresh,
Zoom, etc.

Displays:
3D, 2D, XY Graphs and Beam Tracking

13Home, Setup and Acquisition Tabs:
Set your capture parameters and get the resulting measurements

MAIN CONTROLSThe upper part of the software includes all the main controls in a ribbon format. These controls are grouped by family:
Capture Controls, File Controls, Buffer Controls, M2 Controls and Data Computations. The last includes very useful filters
and a normalization function.

2

DISPLAYS
T he left-hand side of the software is the display panel. Four displays are available: 3D, 2D, XY (cross-sectional graphs along thecrosshairs) and Beam Tracking. The desired display is selected by clicking on the corresponding icon at the bottom of thepanel. Print screen controls are available for the 3D, 2D and Beam Tracking displays. They allow the user to save an image ofthe current view in BMP format.

3

HOME AND SETUP TABS
The right-hand side of the software contains the Home, Setup and Data Acquisition tabs. The Home tab allows the user toselect the main controls for his measurements (Beam Diameter Definition, Crosshair Center and Orientation) and displays
the resulting measurements below. The Setup tab allows the user to set the measurement parameters (Exposure Time,
Image Orientation and Averaging, Active Area, etc.) and the Data Acquisition tab allows the user to save measurementswith or without full images, to enter the Sampling Rates and a Total Duration for the Acquisition. More tabs with advanced
controls are available when clicking on the Show/Hide Options button in the Computations panel.

110

gentec-eo.com/laser-beam-diagnostics

BEAMAGE
Software features

3D DISPLAY

Reset View

Top View

ENERGY DETECTORS

Print Screen

2D DISPLAY

Diameter

SelectActive Area

Print
ScreenReset View

Show/Hide
Diameter

BEAM PROFILINGThe 2D display features the crosshairs (set to the major and minor axis or along a
specific angle) and the measured diameters of the beam. These diameters vary withthe chosen definition (4-sigma, FWHM, 1/e2, etc.) and the display can be turned ON
or OFF. The Print Screen button allows to save a picture of the current screen in BMP
format.

Crosshairs

POWER DETECTORSThe 3D display shows the actual shape of the beam. It is possible to easily zoom, pan
and rotate the image. The Reset button puts the data back in its original configuration.
This display also features a Print Screen button to save the latest image in BMP format.

Toggle tograyscale

XY DISPLAY

Zoom

Gaussian Fit

Show/Hide
CursorShow/Hide
FWHM

Show/Hide1/e2

Semi-Log Graph

BEAM TRACKING DISPLAY
Mean: -20.427, 7.319
Last: -205.513, 624.418

Azimut: 0.006° ∆x: 1416.259∆: 152.890∆y: 459.746

0 um-2992 um
-5632 um
Number of samples : 2000/2000

PRODUCT GUIDE 2024

0 um

Save Data

Print Screen

Zoom

Pick Origin Point

Reset View

5632 um

T 418.651.8003 | info@gentec-eo.com.com

CUSTOM / OEM PRODUCTS

The Beam Tracking Display allows the user to visualize the variation of the beam’s
centroid position on the sensor. This display shows the latest calculated position as wellas the previous ones, until the user resets the view. The display also shows the mean
position of all computed values and gives information regarding position stability forboth X and Y axes. This tool is great to monitor the beam pointing stability over time.

2992 um

DISPLAYS & PC INTERFACES

Save Data

TERAHERTZ DETECTORSThe XY display plots cross-sectional graphs of the beam along the crosshairs. This
display features many useful tools like zoom, cursor, and FWHM and 1/e2 level bars. It isalso possible to display the graphs in semi-log format to enhance the details in the low
intensity parts of the beam.

111

BEAMAGE-M2
Automated M2 measurement system

AUTOMATED MEASUREMENTS

Inside the BEAMAGE-M2, a computer-controlled motorized rail allows
precise positioning of two mirrors, which in turn allow a 400 mm beam path
difference. At each position of the translation stage, a beam profile is acquired
and the beam diameter is measured. The automation of the translation stage
allowed by the software is the key to a fast measurement.

KEY FEATURES

PRACTICAL ALIGNMENT TOOLLARGE APERTURESThe only M2 system on the market equipped with acomplete set of 50mm (2’’) optics. Also, the sensor
is 11.3 x 11.3mm
SIMPLE ALIGNMENT
Two beam-steering mirrors are included for quick
and easy alignment of your laser into the system.
COMPACTThe low-profile ingenious mechanics make it easy
to fit the device on any optical table
ISO COMPLIANT
The calculations are fully compliant to the ISO
11146 and 13694 standards
FAST ACQUISITION
Make a complete, ISO-compliant measurement in
only 20 seconds with the ROI feature and in lessthan a minute with full-frame acquisition
FLEXIBLE & INTUITIVE SOFTWARE

112

gentec-eo.com/laser-beam-diagnostics

Each BEAMAGE-M2 system includes an alignment tube that helps you set up
the system faster. Simply use the two alignment mirrors to center your laser
beam onto both irises, and you will be ready to start measuring in no time!
The fluorescent material around the pinholes also helps to align beams that arein the NIR range without having to use an IR viewer.

BEAMAGE-M2
Specifications

POWER DETECTORSBEAMAGE-M2
SENSOR TECHNOLOGY

Beamage-4M included

EFFECTIVE APERTURE

Ø 48 mm optics - 11.3 x 11.3 mm sensor

MEASUREMENT CAPABILITY
System wavelength range

350 - 1100 nm

Attenuation range

ENERGY DETECTORS

Beam diameter range

3 Flip-mount attenuators for 8 levels of attenuation: no attenuation, ND0.5, ND1, ND2, ND1.5, ND2.5, ND3, ND3.5a

55 µm to 11.3/3 mm

Translation stage
Mechanical travel range

200 mmEffective optical path range

400 mm
5 AR-coated lenses included: 200 mm, 250 mm, 300 mm, 400 mm and 500 mm

Typical M2 accuracy b± 5%

Typical M2 repeatability b± 2%

Applicable light sources

CW and quasi-CW

Typical measurement time

45 s with full-frame acquisition

DAMAGE THRESHOLDS c
Maximum average power
Maximum density (1064 nm)

BEAM PROFILING

Lens focal length

1 W with ND filter
CW: 10 W/cm2 ; Pulsed: 0.1 J/cm2

PHYSICAL CHARACTERISTICS
Dimensions
357 mm (L) x 165 mm (W) x 135 mm (H)

Total (including external mirrors)

602 mm (L) x 193 mm (W) x 172 mm (H)

Optical axis height

86 mm

Weight6.6 kgPower supply

48 VDC, 1.25A out

SOFTWARE
Displays

TERAHERTZ DETECTORSMain enclosure

2D, 3D, XY, Beam Tracking and M2
D4σ
1/e2 along crosshairs (13.5 %)FWHM along crosshairs (50 %)Custom ( %)Beam quality definitions

Laser beam quality M2: M2x, M2y (ISO compliant)Beam Propagation Factor: BPPx, BPPy
Width at waist: Wx, Wy
Waist location and offset: Zx, Zy, ΔZ
Divergence angle: θx, θyRayleigh length: ZRx, ZRy
Astigmatism

Printing and reports

Full report in print-ready format

DISPLAYS & PC INTERFACES

Beam diameter definitions

CUSTOM / OEM PRODUCTS

ORDERING INFORMATION
Product page

Specifications in the table above are for the use with a Beamage-4M beam profiler (included in the Beamage-M2 kit)a. At the Beamage sensor
b. Depending on the beam quality and optical configurationc. With ND4 filter at the Beamage
PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

113

BEAMAGE-M2
Software features

Select which set of Rayleigh range boundaries
to display on the graph: X, Y or both

Switch to 3D or 2D displays to seeeach of the measured profiles

114

gentec-eo.com/laser-beam-diagnostics

M2 Results tab:
View and understand all the measured parameters
quickly, for both the initial laser beam and thebeam inside the BEAMAGE-M2 system

M2 Setup tab:Control your acquisition parameters

BEAMAGE-M2
Software features

ISO COMPLIANCE MADE SIMPLE

The automatic settings are updated after each calculation, considering the values of Z0
and ZR from the latest measurement.

POWER DETECTORSWith the “RUN M2 SETUP” button, the software automatically defines new parameters fora more precise M2 measurement. The “ISO SCAN” data set complies with the ISO-11146 M²
measurement standard, being spread between -3ZR and +3ZR.

By default, the results graph always shows the calculated positions of the first three
Rayleigh distances on each side of the waist.
ENERGY DETECTORSBEAM PROFILING

FULL CONTROL ON YOUR DATA
During an M2 scan, each of the measured profiles is saved and the flexible software
gives you complete control on your acquired data.
• View each acquired profile in 2D display or 3D display.
• Add measurement points to a data set at the position of your choice with the “ADD”
button.

TERAHERTZ DETECTORS• Remove unwanted profiles from your data set & recalculate the measurements.

DISPLAYS & PC INTERFACESFAST ATTENUATION
Add or remove attenuation with the flick of a finger. The software adjusts the exposure
time at each frame during an acquisition, and advises the user on the required
attenuation.

CUSTOM / OEM PRODUCTS

PRODUCT GUIDE 2024

T 418.651.8003 | info@gentec-eo.com

.com

115

ND

Neutral density (ND) filters - up to 1 W

COMPATIBLE PRODUCTS

BEAMAGE-4M

KEY FEATURESFOR ALL BEAMAGE MODELS
We offer various SM1-threaded absorptive ND filtersthat can be fixed directly on the aperture of the
Beamage camera via a SM1 to C-mount adaptor.
We also offer SM2-threaded filters that can be fixed
on the Beamage-4M-FOCUS via a SM2 to T-Mountadaptor.
HIGH-QUALITY OPTICS
These filters reduce the intensity of all wavelengths
without affecting the wavefront of the beam or
distorting the image.
STACKABLE ATTENUATION
Subsequent filters can be stacked directly on eachother. Sets of 3 filters or 6 filters as well as individual
filters are available.

116

gentec-eo.com/laser-beam-diagnostics

BEAMAGE-4M-FOCUS

PH series

OVERVIEW OF THE MODELS
MODEL NAME

EQUIVALENTATTENUATION

TRANSMITTANCE
AT 633 NM

SM1 FILTERS
ND0.5

SM2 FILTERS
ND0.5-FOCUS(1/3.16)

~32 %

ND1.0

ND1.0-FOCUS(1/10)

~10 %

ND2.0

ND2.0-FOCUS(1/100)

~1 %

ND3.0

ND3.0-FOCUS(1/1000)~0.1 %

ND4.0

ND4.0-FOCUS(1/10 000)

~0.01 %

ND5.0

ND5.0-FOCUS(1/100 000)

~0.001 %NDSET-6 (Set of 6 filters)
NDSET-3 (Set of 3 filters: ND1, ND2, ND3)
ND-H (ND filter holder)

ND

Specifications

POWER DETECTORS

ND0.5 TO ND5.0
400 nm a - 1595 nm

Filter diameter

25 mm ØClear aperture

22.5 mm Ø (90 % of diameter)

Dimensional tolerance+0.0/-0.25 mm

Optical density tolerance± 5%

Parallelism< 10 arcsec

Transmitted wavefront error< λ/10 at 633 nm

Surface flatness< λ/4

Surface quality

40 - 20 Scratch-Dig

Maximum power

1W

Damage thresholds

100 W/cm2 or 3 J/cm2

ENERGY DETECTORS

Spectral range

Product page

BEAM PROFILING* Data specified at 633 nm
a. For ND4.0 filter, lower limit with other models.

SPECTRAL TRANSMISSION OF ALL FILTERSTERAHERTZ DETECTORS

100
ND0.5
ND1.0
ND2.0

10

ND3.0

1

DISPLAYS & PC INTERFACES

Transmission (%)

ND4.0
ND5.0

0.1

CUSTOM / OEM PRODUCTS

0.01

0.001
200

400

600

800

1000

1200

1400

1600

Wavelength (nm)PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

117

BA

Optical attenuators - up to 1 kW

MULTIPLES USES
Monitor power and beam profile simultaneously
Polarization insensitive beam-splitter with no back-reflections
Optical pick-off for use with our energy or power detectors
Attenuator for our high sensitivity detectors like M6 series and PH series

COMPATIBLE PRODUCTS

KEY FEATURESMANAGE THE LASER POWER
CMOS sensors have low saturation levels as well as low
damage thresholds. It is thus very important that you
control your laser power to get the best measurement
possible and avoid damaging the BEAMAGE camera.
SAMPLE YOUR LASER BEAM
The BA series optical attenuators use Fresnel
reflections on two orthogonal wedges to pick off asmall fraction of the input beam. The incoming beampolarization state and irradiance are preserved.
IMPROVED MECHANICS
The BA16 models are now compatible with 30-mm
cage systems and also include SM1 threads on the
input face

118

gentec-eo.com/laser-beam-diagnostics

BEAMAGE-4M

BEAMAGE-4M-FOCUS(for BA32-1KW only)

PH series

M6 series

UP55N-40S-H9(for BA32-1KW only)

ACCESSORIESBA32 mounting kit for
BEAMAGE-4M

BA32 mounting kit for
BEAMAGE-4M-FOCUS

BA32 mounting kit for
UP55N-40S-H9

BASpecifications

POWER DETECTORS

BA16-60S

BA16K-150S-H5-D0

BA16K-500F-H9-D0

BA32-1KW

MAXIMUM POWER

60 W

150 W

500 W

1000 W

EFFECTIVE APERTURE

16 mm Ø16 mm Ø16 mm Ø32 mm ØCOOLING METHOD

Convection

Convection

Fan

Water

Spectral range

200 - 2100 nm

200 - 2100 nm

200 - 2100 nm

Integrated power detector

N/A

UP19K-15S-H5-D0

UP19K-110F-H5-D0

200 - 2100 nm
Compatible with UP55 series

Fan input voltage

N/A

N/A

12 VDC

N/A

Equivalent attenuation

1/1700 @ 1064 nm

1/1700 @ 1064 nm

1/1700 @ 1064 nm

1/1900 @ 1064 nm

Optical wedges material

UV fused silica (uncoated)

UV fused silica (uncoated)

UV fused silica (uncoated)

UV fused silica (uncoated)

Residual beam deviation

5.6°

5.6°

5.6°

3.6° @ 1064 nm

Polarization correction

Yes (pair of orthogonal wedges) Yes (pair of orthogonal wedges) Yes (pair of orthogonal wedges) Yes (pair of orthogonal wedges)

MEASUREMENT CAPABILITY

Maximum energy density

150 W

500 W

1000 W

10 MW/cm2

10 MW/cm2

10 MW/cm2

10 MW/cm2

10 J/cm2

10 J/cm2

10 J/cm2

10 J/cm2

16 mm Ø16 mm Ø16 mm Ø32 mm ØDimensions
Weight45H x 47W x 81D mm
0.26 kg

54H x 50W x 91D mm
0.37 kg

54H x 54H x 126D mm
0.46 kg

145H x 250W x 132D mm
5.5 kgMounting thread

SM1

SM1

SM1

SM2

Included adaptor

SM1 external threaded tube

SM1 external threaded tube

SM1 external threaded tube

N/A

STAND-S-233

STAND-S-233

STAND-S-233

2x STAND-S-443-C-M

BEAM PROFILINGPHYSICAL CHARACTERISTICS
Aperture diameter

60 W

ENERGY DETECTORSDAMAGE THRESHOLDS
Maximum power
Maximum average power density(not included)

ORDERING INFORMATION
Compatible StandTERAHERTZ DETECTORS

Product PageDISPLAYS & PC INTERFACESCUSTOM / OEM PRODUCTS

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

119

UV CONVERTERS
Wavelength management

SPECIFICATIONSBSF23C11.3N BSF23P11.3N BSF23R11.3N BSF23G11.3N
Input Aperture Ø

23 mm

23 mm

23 mm

23 mm

Overall Length (OAL) 97 mm

97 mm

97 mm

97 mm

Magnification

1.4

1.4

1.4

1.4

Crystal Type

C

P

R

G

Wavelength range
Saturation level
193 nm
248 nm
308 nm110 - 225 nm

10 - 350 nm

110 - 532 nm

X-ray - 400 nm

400 mJ/cm2

30 mJ/cm2

50 mJ/cm2

10 mJ/cm2

N/A
N/A

30 mJ/cm
50 mJ/cm2

400 mJ/cm
400 mJ/cm2

10 mJ/cm2
50 mJ/cm2

Decay time3 - 5 μs

5 μs

4000 μs

0.1 μsMax repetition rate

20 - 30 kHz

20 kHz

25 Hz

20 kHz22

Product page

A complete procedure on how to choose the appropriate UV Converter (UV
Converter Application Note) is available on our website at www.gentec-eo.com.
UV Converters take advantage of a phenomenon
called fluorescence to extend the performance rangeof the Beamage beam profiling camera to ultraviolet
wavelengths. A fluorescent crystal located at theentrance of the converter absorbs UV wavelengths
and reemits longer wavelengths (in the visible
spectrum), which are less energetic and detected by
the CMOS sensor.

MAIN CHARACTERISTICS
Transforms wavelengths contained between X-rays and
400 nm to visible and near-IR wavelengths.
Images larger beams due to the magnification
properties of the optics.
Built with an iris at the output port for a control of the
exposure on the CMOS sensor.
Removable extension tube that is easily fixed onto theentrance port of the Beamage camera.
Ready to use within minutes

120

gentec-eo.com/laser-beam-diagnostics

IR ADAPTOR
Wavelength management

SPECIFICATIONSPOWER DETECTORSIR ADAPTOR
27.5 mm Ø

IR spectral range

1495 nm - 1595 nm

Peak IR sensitivity

1510 nm and 1540 nm

Converted wavelengths

950 nm - 1075 nm

Pixel Multiplication Factor

3.29

Minimum beam size

230 µmMaximum beam size

19 mm

Maximum resolution

12 lp/mm over active area 40 lp/mm at sensor focal plane

Distortion-1.0 % barrel distortion (inverted image)Linearity

Non-Linear, IR converted output ~ IR input intensity 1.41

Spectral transmission

360 nm - 2000 nm at F30.8

Damage threshold

1 W/cm2

Dimensions

46 mm Ø x 97 mm L

Operating temperature-10°C to +40°C

Weight210 g

ENERGY DETECTORS

Active area

BEAM PROFILING

Product page

MAIN CHARACTERISTICS

EXCITATION SPECTRUM

Removable and easily C-mounted onto the entrance
port of the camera.

6
5
4
3
2

CUSTOM / OEM PRODUCTS

Built with a high quality coated anti-reflection input
window that allows wavelength conversion with low
distortion and maximum image resolution.

7
Signal intensity (a.u.)

Images larger beams due to the convergent
properties of the optics (3.29X).

DISPLAYS & PC INTERFACES

Converts wavelengths between 1495 nm and 1595
nm to shorter wavelengths between 950 nm and1075 nm.

Ready to use within minutes.

TERAHERTZ DETECTORS

Typically, a CMOS silicon sensor is operating at its
full potential when imaging lasers with wavelengths
between 350 nm and 1150 nm. If you want toextend the performance range of your Beamage
beam profiling camera to the near-IR telecom
wavelengths band, you can use the IR Adaptor. This
ideal solution takes advantage of a multi-photon
absorption process to extend the sensitivity rangeof the camera sensor to a portion of the near-IR
spectrum (1495 nm - 1595 nm).

1
0

1500

1550

1600

Wavelength (nm)PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

121

FILTERS

Wavelength management

UV BANDPASS FILTER
We also offer a color glass filter specially designed for the UV spectrum. Depending on the wavelength, the UG11-UV filter transmits
20 % to 70 % of the input beam power. It is particularly useful for applications with wavelengths contained between 250 nm and
370 nm. Other wavelengths are blocked by the filter.

SPECIFICATIONSSPECTRAL TRANSMISSION
UG11-UV90%

250 nm - 370 nm

80%

Diameter

25 mm Ø70%

Clear aperture

80 % of area

60%

Dimensional tolerance+0.0 / -0.2 mm

50%Thickness

3 mm

Thickness tolerance+0.0 / -0.2 mm

Parallelism< 3 arcmin

Surface flatness< λ/4

Maximum power

1W

Surface quality

40 - 20 Scratch-Dig

Damage threshold

30 W/cm2 (typical)

Relative Response (%)

MODELSpectral range

40%
30%
20%
10%
0%
-10%

200

300

400

500

600

700

800

900

1000

1100

Wavelength (nm)

Product page

* Data specified at 633 nm

IR FILTER
The B3-IR-FILTER is a color glass filter specifically designed for IR applications. Acting as a longpass filter, the B3-IR-FILTER cuts all
the wavelengths below 1250 nm and only lets the IR wavelengths pass. It transmits approximately 70 % of the incident light.

SPECIFICATIONSSPECTRAL TRANSMISSION
B3-IR-FILTER

100%

Spectral range

1250 - 1350 nm

90%

Diameter

25 mm Ø80%

Clear aperture

80 % of area

70%

Dimensional tolerance+0.0/-0.2 mm

Thickness

6.3 mm max

Parallelism< 3 arcmin

Surface flatness< λ/4

Maximum power

1W

Surface quality

80-50 Scratch-Dig

Damage threshold

30 W/cm2 (Typical)

Product page

122

gentec-eo.com/laser-beam-diagnostics

Relative Response (%)

MODEL

60%
50%
40%
30%
20%
10%
0%

350

550

750

950
Wavelength (nm)

1150

1350

CL

Beam size management

CAMERA LENSES
It is necessary to use a camera lens to image beams that are larger than the CMOS sensor (11.3 mm X 11.3 mm) of the Beamage beamprofiling camera. A camera lens can be directly C-mounted onto the aperture of the Beamage camera.

SPECIFICATIONS
CL-50

25 mm

50 mm

Horizontal FOV

14°

7°

FOV at 1m

245 mm

120 mmMinimum working distance

0.5 m

1m

Maximum beam size

2000 mm X 2000 mm(not a limiting factor)

2000 mm X 2000 mm(not a limiting factor)

mechanism

Very high because of indirect
mechanism

Inverted image

Yes

Yes

Beam distortion

Setup, lens aberration and speckles
from diffusing glass

Setup, lens aberration and speckles
from diffusing glass

Diffusing material needed

Yes

Yes

Magnification calibration needed

Yes

Yes

Possibility of wavelength conversion

Yes

Yes

Optical filter needed

Rarely to neverRarely to never

Removable

Yes

Yes

BEAM PROFILING

Maximum measurable intensity / energy Very high because of indirect

ENERGY DETECTORS

CL-25
Focal length

POWER DETECTORS

Camera lenses work by indirectly imaging on the sensor the reflection or the transmission of a beam that previously went through adiffusing material such as glass (see diagrams below).

Product page
TERAHERTZ DETECTORSIMAGING A TRANSMITTED BEAM

IMAGING A REFLECTED BEAMDISPLAYS & PC INTERFACES

BeamageDiffuser

Gentec-EO detector
or beam dump

Beamage
Camera Lens

Camera Lens
Sensor Plane

Laser

Laser

CUSTOM / OEM PRODUCTS

Sensor
Plane

Image ofthe Beam

Image ofthe Beam

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

123

ABSORPTION CURVES
Specifications

Beamage relative response

100%
90%

Relative Sensitivity (a.u.)

80%
70%
60%
50%

Beamage-4M-FOCUS

40%

Beamage-4M

30%
20%
10%
0%

350

450

550

650

750

850

Wavelength (nm)

124

gentec-eo.com/laser-beam-diagnostics

950

1050

1150

ABSORPTION CURVES
Specifications

Beamage operating range

532 nm632 nm

800 nmPOWER DETECTORS

355 nm100,000

1064 nm 1150 nm

ENERGY DETECTORS

10,000BEAMAGE-4M-FOCUS
BEAMAGE-4M
100

BEAM PROFILING

Power Density (mW/cm2)

1,000

Saturation Level(With ND4.0 Filter, 0.06 msec Exposure Time)

101

TERAHERTZ DETECTORS

0.1
BEAMAGE-4M

BEAMAGE-4M-FOCUS

0.01
Minimum Measurable Power450

Wavelength (nm)

550

650

750

850

950

1050

DISPLAYS & PC INTERFACES

0.001
350

(With ND4.0 Filter, 200 msec Exposure Time)

1150

Never exceed 1 W of Average Power on the Filter

CUSTOM / OEM PRODUCTS

PRODUCT GUIDE 2024

T 418.651.8003 | info@gentec-eo.com

.com

125

TERAHERTZ DETECTORS
Overview of the different models

WHAT IS TERAHERTZ RADIATION?
The THz portion of the electromagnetic spectrum fills the gap between the far infrared and the microwaves. More precisely, it is nestled
between the high-frequency edge of the microwave band, 300 GHz (3×1011 Hz), and the long-wavelength edge of far-infrared light,
3000 GHz (3×1012 Hz or 3 THz). In wavelengths, this range corresponds to 0.1 mm (or 100 μm) infrared to 1.0 mm microwave. The THzband is set in the region where electromagnetic physics can best be described by its wave-like characteristics (microwave) and its
particle-like characteristics (infrared).

Electronics
Frequency

10-3 THz
Radiowaves

Wavelength

30 cm

10-2 THz

THz Gap
0.1 THz

1 THz

Photonics
10 THzMicrowaves

3 cm

3 mm

100 THzIR

300 µm30 µm1000 THz 10000 THz
Vis

3 µm

UV

300 nmX-Rays

30 nm

WHAT IS IT USED FOR?

HOW IS IT MEASURED?

THz radiation is interesting because of the way it interacts withmatter:THz sources range in power from nW to mW and in energy
from nJ to mJ. Like most electromagnetic sources, they must
be characterized for performance and/or control.

• It can penetrate things like wood, plastics, clothing, and othermaterials.

Older THz detection methods include:• It is also absorbed by water, or a material that contains water, likehuman skin.

• Golay cells• It is non-ionizing and therefore not harmful to humans like X-rays
can be.

• Electronic antennas

One of the first uses is the “full body scan” used at airports. It also
has uses in medical applications for early detection of cancer cells.

• Microbolometers
Newer THz detection methods include:• Pyroelectric detectors• Schottky diode detectors• Photoacoustic detectors

WHY ARE GENTEC-EO PRODUCTS BETTER?
Golay cells are large, fragile, costly and have a limited measurement range.
Pyroelectric detectors (like the ones used in our THZ detectors) are small, sensitive, durable and less costly. Some of their advantages are:• High performance in a small package• Broad spectral response (from 0.25 to 3000 µm)• Wide dynamic range (from nW to mW)• Rugged and durable• Very cost-effective

126

gentec-eo.com/terahertz-power-meter

TERAHERTZ DETECTORS
Overview of the different models

POWER DETECTORS

We have designed a unique line of thermal sensors and meters for measurements in the THz region. These products are used
to measure power (radiant flux or irradiance) and energy of CW, pulsed and quasi-CW THz sources from 30 THz to 0.1 THz. Our
product line includes the QS series of miniature pyroelectric detector hybrids that include an operational amplifier and can be easily
integrated into THz instrumentation.

THZ-B DETECTORS, WITH T-RAD RADIOMETERS• Large area: 5 mm and 9 mm Ø
• Wide dynamic range: 10 nW to 20 mW• Broad spectral response with BL coating: 0.1 μm to 3000 μm• Two output options:• “DA” models used with our T-RAD-ANALOG power supply, for analog BNC output to youroscilloscope or lock-in amplifier.

C HOICE OF ANALOG OR DIGITAL OUTPUT

ENERGY DETECTORS• “DZ” models used with our digital T-RAD lock-in radiometer, for USB output to PC.
Includes a powerful standalone LabVIEW software.

THZ5I-BNC: ANALOG RADIOMETER & JOULEMETER
BEAM PROFILING• Very sensitive pyroelectric radiometer and joulemeter• 5 mm Ø detector with BL broadband absorber• Wide dynamic range: 10 nW to 63 μW and 10 nJ to 2 μJ
• Analog BNC output to your oscilloscope or lock-in amplifier

C AN BE USED BOTH FOR POWER AND ENERGY MEASUREMENTS

Available with

• THZ12D-3S-VP:• Large area: 12 mm Ø

• Wide dynamic range: 0.1 mW to 3 W• Spectrally flat (± 5%) from 30 THz to 0.7 THz• Large area: 9 mm Ø

• High sensitivity: 500 nW to 20 mW• Spectral correction factors provided from 0.25 μm to 3000 μm

COMPATIBLE WITH OUR STANDARD DISPLAYS & PC INTERFACES

QS-THZ: MINIATURE HYBRID PYROELECTRIC DETECTORSCUSTOM / OEM PRODUCTS• Standard sizes: 5 and 9 mm Ø active area (other sizes available)• Wide dynamic range: nW to mW and nJ to mJ• Broad spectral response with BL coating: 0.1 μm to 3000 μm• Small packages: TO5 & TO8• Includes integrated operational amplifier

TRY THEM WITH OUR EVALUATION TEST BOX QS-I-TEST
PRODUCT GUIDE 2024

T 418.651.8003 | info@gentec-eo.com

DISPLAYS & PC INTERFACES• THZ9D-20mS-BL:

TERAHERTZ DETECTORS

THZ-D: POWER DETECTORS FOR USE WITH GENTEC-EO METERS

.com

127

THZ-B

THz-B detectors and T-Rad modules

OUTPUT OPTIONSSMART DB15 CONNECTOR
Contains all the calibration data
TWO DETECTOR TYPES AVAILABLE:"DZ": digital output, used with T-RAD digital power module"DA": analog output, used with T-RAD-ANALOG analog power module

COMPATIBLE DISPLAYS & PC INTERFACES

T-Rad Module

KEY FEATURES

T-RAD(for "-DZ" models only)

T-RAD-ANALOG(for "-DA" models only)

COVERS THE ENTIRE THZ SPECTRUM
Get the best precision across the entire wavelength range
and relative measurements from 30 THz to 0.1 THz.
ROOM TEMPERATURE OPERATION
Easier to use and less expensive than a Golay cell.

ACCESSORIESMEASURE POWER FROM nW TO mW
With state of the art pyroelectric sensors, measuredown to 10 nW with 1 nW NEP	USE WITH T-RAD THZ MODULE OR
T-RAD-ANALOG POWER MODULE
Each head can be connected to an oscilloscope using
the analog power module (T-Rad-Analog) or directly toa PC with the digital power module (T-Rad)Stand with delrin post

Removable IR windows(Various types available)

Winston cone

Pelican carrying case

SEVERAL SENSOR SIZES AVAILABLE
Choice of 5 mm and 9 mm diameter
CALIBRATED AT 0.63 µm
All THz detectors are calibrated at a single wavelength(0.63 µm) and include a typical wavelength correction
data from 0.25 to 440 µm. They are used for relative
measurements outside that range.
SDC-500 OPTICAL CHOPPER
All THZ-B detectors require the use of an opticalchopper, like our SDC-500, to sync the signal at either
5 Hz (DA models) or 25 Hz (DZ models)128

gentec-eo.com/terahertz-power-meter

SDC-500 digital
optical chopper

THZ-B
Specifications

THZ5B-BL-DA

THZ9B-BL-DZ

THZ9B-BL-DA

MAX AVERAGE POWER

20 mW

25 μW

20 mW

125 μW

EFFECTIVE APERTURE

5 mm Ø

5 mm Ø

9 mm Ø

9 mm Ø

COMPATIBLE MODULES

T-Rad

T-RAD-ANALOG

T-Rad

T-RAD-ANALOG

Frequency

0.1 - 30 THz

0.1 - 30 THz

0.1 - 30 THz

0.1 - 30 THz

Wavelength

3000 - 10 µm

3000 - 10 µm

3000 - 10 µm

3000 - 10 µm

Max measurable power

20 mW

25 µW

20 mW

125 µW

Noise equivalent power (NEP)

5 nW

1 nW

50 nW

3 nW

Rise time (0-95 %)≤ 0.2s≤ 0.2s≤ 0.2s≤ 0.2sSensitivity (Typical)

N/A

140 kV/W

N/A

20 kV/W

Chopping frequency b

25 Hz

5 Hz

25 Hz

5 Hz

10 mW/cm2

10 mW/cm2

10 mW/cm2

10 mW/cm2

Effective aperture

5 mm Ø

5 mm Ø

9 mm Ø

9 mm Ø

SensorPyroelectric

Pyroelectric

Pyroelectric

Pyroelectric

Absorber

BL

BL

BL

BL

Dimensions

66.0Ø x 46.5D mm

66.0Ø x 46.5D mm

66.0Ø x 46.5D mm

66.0Ø x 46.5D mm

Weight

227 g

227 g

227 g

227 g

STAND-D-233

STAND-D-233

STAND-D-233

STAND-D-233

POWER DETECTORS

THZ5B-BL-DZ

MEASUREMENT CAPABILITY
Spectral range a
ENERGY DETECTORSDAMAGE THRESHOLDSBEAM PROFILING

Max average power density (at 1064 nm)
PHYSICAL CHARACTERISTICSTERAHERTZ DETECTORSORDERING INFORMATION
Compatible standProduct pageDISPLAYS & PC INTERFACES

a. Projected spectral range.
From 10 to 440 μm, spectrometer measurement.
From 440 to 3000 μm, relative measurement only.
This spectral range is subject to change.
b. SDC-500 digital optical chopper sold separately.

CUSTOM / OEM PRODUCTS

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

129

THZ-B
Specifications

T-Rad
The T-Rad is a microprocessor-based digital radiometer that includes a 12-bit ADC
and unique DSP Lock-In Software. It is powered by a USB connection, which also
acts as a Virtual COM port. When a THZ-B Terahertz Pyroelectric detector is plugged
into the T-Rad module, the module reads the content of the head’s EEPROM,
which identifies the detector and provides calibration and wavelength correction
data. The LabVIEW Software supplied with the device makes it very easy to set up
the radiometer, measure a THz or broadband source and record data. The software
is compatible with Windows 7, 8 & 10.

SPECIFICATIONS & FEATURES
T-Rad(rear view)

T-Rad-Analog(front view)

T-RAD

T-RAD-ANALOG

Compatible detector heads

THZ-B-DZ

THZ-B-DA

Full scale ranges

200 nW - 200 mW*

N/A

Power on light

Green

Green

Analog output

0 to 3.6V, BNC± 4.88 V, BNC

PC connection

USB 2.0

None

Trigger input (TTL)

BNC connector

None

Power supply

USB 2.0201849Product number* Actual ranges vary based on the THZ-B detector selected

External, 100/240 VAC 50 - 60 Hz, and 9V battery(both included)202306

INSTRUMENT CONTROL AND STRIP CHART
Instrument controls and the radiant power measurement are always visible,
making it easy to change the radiometer settings, no matter which display
tab is selected. Instrument controls include: Range, Filter Tau, Batch Size, Data
Collection Mode, Reset Options, and a Null button for background cancellation.
In addition, there are more set up and operation status indicators including:detector Rv, Wavelength, Frequency (actual), Locked and Frequency in Range
lights. The Strip Chart displays the Radiant Power measurement in Watts, either
continuously or by the batch. Select full scale, auto scale or use our manual
scaling option.

INSTRUMENT CONTROL AND TUNING NEEDLE
The “TUNE” tab selects the very useful “Tuning Needle” display. This is asimulated analog meter whose speed is determined by the “filter tau” setting.
It is expected to be used during the set-up of a radiometer with a source. The“tau” value is usually set to a small value when aligning the probe to the source(i.e. when peaking the reading). There is a button control to select “full scale”,“min-max” or “reset”. In the “min-max” mode, the indicators are “blue” for theminimum power and “red” for the maximum power.

130

gentec-eo.com/terahertz-power-meter

THZ-B
Specifications

INSTRUMENT CONTROL AND STATISTICS

THZ-B
Detector

Oscilloscope

THZ-B-DA DETECTOR AND OSCILLOSCOPE

ENERGY DETECTORS

SDC-500
Chopper

POWER DETECTORS

THz
SourceIn the “Statistics” tab there are 4 large windows that contain the statistics for theselected batch, including: Minimum, Maximum, Standard Deviation and Mean,
expressed in Engineering Notation. Standard Deviation can be displayed in Watts
or as a % that is user-selectable. There is also a window that shows the bandwidthof the Digital Band Pass Filter based on the user selected “Filter Tau” (0.100 to 100
seconds). A lower time constant is helpful when setting up, and a longer one when
making measurements, especially on the lower ranges of the instrument.

T-Rad-Analog
Power Module

THz
SourceSDC-500
Chopper

THZ-B
Detector

Lock-In Amplifier

Chopper
Ref Input

T-Rad-Analog
Power Module

THZ-B-DZ DETECTOR AND T-Rad MODULE
THz
SourceSDC-500
Chopper

THZ-B
Detector

PRODUCT GUIDE 2024

Although analog solutions are available, for simplicity, convenience and sensitivity,
we recommend you choose our THZ-B-DZ detectors and the T-Rad Digital
Radiometer. Our unique DSP Lock-In Amplifier software provides a function much
like the Analog Lock-In, but is so much easier to use. It also addresses thermal driftof the sensor and allows you to display the power measurement and complete
statistics directly in digital and graphic formats. Set the range, null the background,
set the filter tau (bandwidth) and make the measurement. It’s that easy!

T 418.651.8003 | info@gentec-eo.com.com

CUSTOM / OEM PRODUCTS

T-Rad-USB
ModuleComputerDISPLAYS & PC INTERFACES

This is another analog set-up option that we recommend if you have to measure very
low power levels (i.e. less than 5 μW) where the signal may be buried in the broadband
noise. The voltage output of the analog THZ-B-DA detector, powered by our T-RadAnalog, is routed to the Lock-In Amplifier input, and the Sync Output of our SDC-500
Chopper is connected to the reference input. The Lock-In Amplifier will lock on the
chopping frequency and you can dial in a long integrating time and measure a very
low RMS voltage. The voltage divided by our Voltage Responsivity (V/W) equals thepower of the source.

TERAHERTZ DETECTORS

THZ-B-DA DETECTOR AND LOCK-IN AMPLIFIER

BEAM PROFILING

Here is a basic analog set up that would be useful if the optical power of the source
was about 5 μW or greater. The output of the THZ5B-BL-DA detector would beapproximately 600 mV at 5 Hz chopping frequency, giving plenty of signal for an
oscilloscope. Simply read the voltage output and divide by the Rv factor (V/W) of thedetector to measure the intensity of the source in Watts. Also consider applying awavelength correction factor under certain circumstances.

131

THZ-I-BNC

THz detectors with integrated analog module

OUTPUT OPTIONSANALOG OUTPUT
Plug the device directly into your oscilloscope or lock-in
amplifier with the BNC output

KEY FEATURESACCESSORIESCOVERS THE ENTIRE THZ SPECTRUM
Measure accurately from 0.25 to 15 μm and from30 THz to 0.1 THz in relative terms
MEASURE POWER FROM nW TO µW
Make low-level measurements with an NEP of 1.0 nW

Stand with delrin post

MEASURE ENERGY FROM nJ TO µJ
Can be used with low repetition rate pulsed THz
sources to measure pulse energy up to 40 Hz
INTEGRATED ANALOG MODULE
Plug the device directly into your oscilloscope or
Lock-In Amplifier
BATTERY OR EXTERNAL POWER
Includes 9V battery and an external power supply
CALIBRATED AT 0.63 µm
All THz detectors are calibrated at a single
wavelength (0.63 µm) and include typical
wavelength correction data from 0.25 to 440 µm.
They are used for relative measurements outsidethat range.
SDC-500 OPTICAL CHOPPER
The THZ-I-BNC models require the use of an
optical chopper, like our SDC-500, running at 5 Hz.

132

gentec-eo.com/terahertz-power-meter

Pelican carrying case

Removable IR Windows(Various types available)

SDC-500 digital
optical chopper

THZ-I-BNC
Specifications

POWER DETECTORS

THZ5I-BL-BNC
MAX AVERAGE POWER

62.5 µW

EFFECTIVE APERTURE

5 mm ØINTEGRATED MODULE

Analog (BNC)

MEASUREMENT CAPABILITY
Spectral range a
0.1 - 30 THz

Wavelength

3000 - 10 µm

Max measurable power

62.5 µW

Noise equivalent power b

1.0 nW

Rise time (0-100 %)≤ 0.2sSensitivity (Typical)

140 kV/W

Chopping frequency

5 Hz (Required)

Calibration uncertainty

Contact us

ENERGY DETECTORS

Frequency

Maximum measurable energy

2 μJ

Noise equivalent energy

1.0 nJ

Minimum pulse width

1.0 μsMaximum repetition rate

40 Hz

BEAM PROFILING

Energy mode

DAMAGE THRESHOLDS
Maximum average power density (1064 nm)

50 mW/cm2

Effective aperture

5 mm Ø

SensorPyroelectric

AbsorberBL

Analog output

0-10 V

Dimensions

81.3Ø X 99.3D mm

Weight

500 g

TERAHERTZ DETECTORSPHYSICAL CHARACTERISTICS

Compatible stand

DISPLAYS & PC INTERFACESORDERING INFORMATIONSTAND-D-233

Product page

PRODUCT GUIDE 2024

CUSTOM / OEM PRODUCTS

a. Projected spectral range.
From 10 to 440 μm, spectrometer measurement.
From 440 to 3000 μm, relative measurement only.
This spectral range is subject to change.
b. At 632 nm and a chopping frequency of 5Hz.

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

133

THZ-DTHz detectors for use with our universal displays & PC interfaces

OUTPUT OPTIONSSMART DB15 CONNECTOR
Contains all the calibration data
ANALOG OUTPUT
When used with APM (D) analog power supply module
ALL-IN-ONE-METER (for THZ12D only)
Connects directly to a PC
Two models available: 				USB output (-INT)
RS-232 output (-IDR)

COMPATIBLE DISPLAYS & PC INTERFACESKEY FEATURES
COVERS THE ENTIRE THZ SPECTRUM
Get the best precision across the entire wavelength
range and relative measurements from 30 THz to
0.1 THz.

MAESTROU-LINK

M-LINK(for THZ12D)

Stand with steel post
(for THZ12D)

Stand with delrin(for THZ9D)

SDC-500 digital
optical chopper

Pelican carrying case

Extension cables(4, 15, 20 or 25 m)

Extra isolation tube

ROOM TEMPERATURE OPERATION
Easier to use and less expensive than a Golay cell.
CALIBRATED AT 10.6 µm
THZ-D detectors are calibrated at a single
wavelength 10.6 um (30 THz) and at 10 Hz chopping
frequency for the THZ9D. Both include typical
wavelength correction data from 10.6 to 440 µm.
They are used for relative measurements outsidethat range.

APM (D)
analog power module(for THZ9D)

ACCESSORIESLARGE AREA
Models range from 9 mm Ø for the THZ9D and
12 mm Ø for the THZ12D.
WIDE RANGE OF MEASUREMENTS
Measure from 100 uW to 3 W of continuous powerwith the THZ12D model, the highest in our terahertz
range of products, and down to 5 uW to 25 mW with
the THZ9D model.
USE WITH A UNIVERSALDISPLAYS & PC INTERFACENo need for an exclusive monitor. These unique THz
detectors work with our display & PC interface.
SDC-500 OPTICAL CHOPPER
The THZ9D model requires the use of an opticalchopper, like our SDC-500, running at 10 Hz.
Fiber adaptors & connectors(FC, ST and SMA)

134

gentec-eo.com/terahertz-power-meter

THZ-D
Specifications

THZ12D-3S-VP

25 mW

3W

EFFECTIVE APERTURE

9 mm Ø

12 mm ØCOMPATIBLE DISPLAYS & PC INTERFACES

MAESTRO, U-LINK & APM (D)

MAESTRO, U-LINK & M-LINK

Frequency

0.1 - 30 THz

0.1 - 30 THz

Wavelength

3000 - 10 µm

3000 - 10 µm

with MAESTRO

20 mW

3W

with U-LINK

25 mW

3W

300 nW

0.5 µW

N/A

50 - 100 µW

N/A

12 µW/°C< 0.2 s3s

POWER DETECTORSTHZ9D-20mS-BL
MAX AVERAGE POWER

MEASUREMENT CAPABILITY
Spectral range a

ENERGY DETECTORS

Maximum average power

Noise equivalent powerb

Minimum measurable powercThermal drift
Rise time (nominal)d

Minimum repetition rate

7 Hz

10 Hz (required)

N/A

Calibration uncertainty g± 5.0 % at 10.6 µm;± 15 % at 10.6 - 440 µm a

± 3.0 % at 10.6 μm± 8.0 % at 10.6 - 300 μm± 15 % at 300 - 440 μm

Repeatability±0.5 %±0.5 %

Maximum average power density h

50 mW/cm2

30 W/cm2

Maximum energy density< 0.1 J/cm2< 1 J/cm2

Effective aperture

9 mm Ø

12 mm ØAbsorber

BL (Black Absorber)

VP (Volume Absorber)

38.1Ø x 79 mm

73H x 73W x 20D mm(72D mm with tube)

91 g

320 g

STAND-D-233 or STAND-D-233-M

STAND-D-233

BEAM PROFILING

1000 Hz

Chopping frequencyf

DAMAGE THRESHOLDSTERAHERTZ DETECTORSPHYSICAL CHARACTERISTICS

Dimensions
Weight (head only)
ORDERING INFORMATIONDISPLAYS & PC INTERFACES

Compatible standProduct page

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

CUSTOM / OEM PRODUCTS

a. From 10 to 440 μm, spectrometer measurement with multiple laser references validation.
From 440 to 600 μm, spectrometer measurement only.
From 600 to 3000 μm, relative measurement only.
This spectral range is subject to change.
b. Nominal value, actual value depends on electrical noise in the measurement system.
c. Actual value depends on ambient conditions and the measurement system.
d. With anticipatione. Maximum output voltage = sensitivity x maximum power.
f. Minimum repetition rate for stable average power measurements.
g. Including linearity with power.
h. At 1064 nm, 1 W CW.

.com

135

QS-THZ

Hybrid THz detectors

PERMANENT IR WINDOW OPTIONSEvery model can be fitted with a permanent IR window
to narrow the wavelength range:
S5/8: sapphire (0.3 - 4.5 and 100 - 1000 µm)
Q5/8: quartz (0.25 - 3.0 and 50 - 1000 µm)
Si5/8: silicon (1.2 - 8.0 and 50 - 1000 µm)* Pictures for indicative purposes only

KEY FEATURESACCESSORIES	RELATIVE MEASUREMENTS FROM 0.1 TO
30 THZ
Broadband, room temperature operation, easier
to use and less expensive than a Golay cell
EASY TO INTEGRATE FORMAT
TO5 and TO8 packages make the QS-THZ
detectors small and easy to integrate in anexisting system

QS-I-Testevaluation test box

SEVERAL SENSOR SIZES AVAILABLE
Choice of 5 and 9 mm Ø
CALIBRATED AT 0.63 µm
All THz detectors are calibrated at a single
wavelength (0.63 µm) and include a typical
wavelength correction data from 0.25 to440 µm. They are used for relative
measurements outside that range.
TEST BOX AVAILABLE
Can be used with our QS-I-TEST test box which
provides mounting and power supply

136

gentec-eo.com/terahertz-power-meter

Pelican carrying case

Permanent IR Windows(Various types available)

SDC-500 digital
optical chopper

QS-THZ
Specifications

QS9-THZ-BL

VOLTAGE RESPONSIVITY

120 kV/W

30 kV/W

EFFECTIVE APERTURE

5 mm Ø

9 mm Ø

PACKAGETO5

TO8Frequency

0.1 - 30 THz

0.1 - 30 THz

Wavelength

3000 - 10 µm

3000 - 10 µm

Max power density

50 mW/cm

50 mW/cm2

Noise equivalent power

1 nW

3 nW

Detectivity

4.108 cm(Hz)½ /W

2.7.108 cm(Hz)½ /W

120 kV/W

30 kV/W

Effective aperture

5 mm Ø

9 mm Ø

Package

TO5

TO8

SensorPyroelectric

Pyroelectric

Absorber

BL

BL

Dimensions (excluding pins)

9.1Ø x 6.4D mm

15.2Ø x 6.4D mm

Weight45 g

45 g

POWER DETECTORS

QS5-THZ-BL

MEASUREMENT CAPABILITY
Spectral range abVoltage responsivityb

ENERGY DETECTORS

2

PHYSICAL CHARACTERISTICS
BEAM PROFILINGORDERING INFORMATION
Product page
TERAHERTZ DETECTORS

a. Projected Spectral Range.
From 10 to 440 μm, spectrometer measurement.
From 440 to 3000 μm, relative measurement only.
This spectral range is subject to change.
b. 630 nm, 5 Hz

DISPLAYS & PC INTERFACES

QS-I-TEST EVALUATION TEST BOX
QS-I-TEST+9V/-9V

Rf resistors

105 - 1010 Ω

Cf compensating

YES

Package

101.6H x 127W x 58.4D

Optical mount¼-20 threaded

Front bezel

SM1 (1.035-40)

Product number

201693CUSTOM / OEM PRODUCTS

Batteries* For details, contact your Gentec-EO representative

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

137

ABSORPTION CURVES
Specifications

VP ABSORBER
100%
90%
80%
70%

Absorption

60%
50%
40%
30%
20%
10%
0%
0

100

200

300

Wavelength (µm)

138

gentec-eo.com/terahertz-power-meter

400

500

600

ABSORPTION CURVES
Specifications

THZ-BL, THZ-I-BL & QS-THZ-BL
POWER DETECTORS

100%
90%
80%

ENERGY DETECTORS

70%

Absorption

60%
50%
40%

BEAM PROFILING

30%
20%
10%

0

100

200

300

400

500

...

TERAHERTZ DETECTORS

0%
3000

Wavelength (µm)

DISPLAYS & PC INTERFACESCUSTOM / OEM PRODUCTS

PRODUCT GUIDE 2024

T 418.651.8003 | info@gentec-eo.com

.com

139

DISPLAYS & PC INTERFACES
Overview of the display modules

We offer four models of meters with display: MIRO ALTITUDE and MAESTRO for both power and energy measurements, as well as
TUNER and UNO for power readings. Connect one of these display devices to your detector and you have a complete laser power orenergy measurement system

MIRO ALTITUDE

MIRO ALTITUDE is Gentec-EO’s flagship product for reading laser power and energy.
It was designed to help engineers and service technicians increase their productivity
thanks to numerous innovative features in both hardware and software. Enter
modern times of laser beam measurement with MIRO ALTITUDE.
Supercharge your productivity with an intuitive user interface, an extra large screen,
tons of connectivity possibilities, 3 convenient display modes, a built-in dataviewer
and a built-in file manager.

P ROFESSIONAL LASER POWER & ENERGY METER

MAESTRO
The MAESTRO power & energy meter is our top of the line display device with an
extra-large 5.6 in color LCD display and fully touchscreen controls. With its uniqueuser interface and faster electronics, it will do more, in less time, and with less effort
than any other meter on the market!

L ASER POWER & ENERGY METER

TUNER
The TUNER power meter display presents both a large LCD display and an ultrafast
needle. It also features min and max holds for both displays, comet tail needle andbar graph function. The TUNER comes in Gentec-EO’s ergonomic design, with a large
LCD display and easy to use direct access keys.

U LTRA-FAST TUNING NEEDLEUNO
The UNO is a simple power meter display, with large contrast fields and direct accessbuttons. Its extremely low power consumption allows it to work on standard alkaline
batteries, making it the display of choice for service technicians working in the field.
With the lowest price for a display meter, the UNO is the perfect choice when looking
for a reliable, entry-level power meter.

E CONOMICAL POWER METER

140

gentec-eo.com/displays-and-pc-interfaces

DISPLAYS & PC INTERFACES
Overview of the PC interfaces

The Gentec-EO PC interfaces come in various sizes and types to cover all applications. We offer models for power or energy
measurement, or both. Most of our PC interfaces are single-channel, and we also offer models with either 2 channels or even up to 4channels.
POWER DETECTORSLINK SERIESThe U-LINK, P-LINK, S-LINK and M-LINK are PC interfaces for our power or energy
detectors and are provided with free software applications.
• U-LINK is a universal power & energy meter that measures ALL detectors in our
product range up to 10 kHz repetition rate. It has a very small footprint.
• P-LINK is a small power meter, available with either a USB or RS-232 connector.
A 4 Channel version is also available.

ENERGY DETECTORS• S-LINK comes with 1 or 2 channels and measures energy detectors at a very fast rate.
It comes with a USB connector, Ethernet also available in option.
• M-LINK is a universal power & energy meter that measures ALL detectors in our
product range and features a unique noise suppression method.

P C-BASED POWER OR ENERGY METERS

BEAM PROFILINGDISPLAY DEVICESMIRO ALTITUDE

MAESTRO

TUNER

UNO

Power measurement

UP, XLP, PH, HP

UP, XLP, PH, HP, UM-B, THZ-D

UP, XLP, PH & HP

UP, XLP, PH & HP

Energy measurement

QE, also UP & XLP in SSE mode

QE, PE, also UP & XLP in SSE mode

N/A

N/A

Display

10in touchscreen

5.6in touchscreen

3.8in LCD, backlit

3.8in LCD

Output2xUSB, USB-C, RS-232, Ethernet

USB, RS-232, Ethernet,
analog output

Analog output

N/A

Data logging

Internal memory and USB key

USB key

N/A

N/A

External trigger

Yes

Yes

N/A

N/A

Number of channels

1

1

1

1

U-LINK

P-LINK

S-LINK

M-LINK

Power measurement

UP, XLP, PH, UM-B, THZ-D

UP, XLP & PH

UP & XLPUP, XLP, PH, THZ-D

Detector Compatibility
TERAHERTZ DETECTORSDISPLAYS & PC INTERFACESPC INTERFACES

Detector Compatibility
QE, PE, also UP & XLP in SSE mode

UP & XLP in SSE mode

QE, PE, also UP & XLP in SSE mode

QE, PE, also UP & XLP in SSE mode

USB, analog output & sync. out.
RS-232 available on certain models

USB & analog output.
RS-232 available on certain models

USB & Ethernet

USB & analog output

External trigger

Yes

N/A

Yes

Yes

Maximum repetition rate

10 kHz/channel

N/A

10 kHz/channel

1 kHz

Number of channels

1

1 or 4, depending on model

1 or 2, depending on model

1

PRODUCT GUIDE 2024

T 418.651.8003 | info@gentec-eo.com.com

CUSTOM / OEM PRODUCTS

Energy measurement
Output141

DISPLAYS & PC INTERFACES
Overview of the PC interfaces

PC INTERFACES
While the vast majority of Gentec-EO detector heads are compatible with the
U-LINK and S-LINK PC interfaces, a few of our specialized detectors requiredifferent data processing methods. In this case, we offer dedicated PC interfacesthat are optimized for these measurements.

H IGH-PERFORMANCE ELECTRONICS FOR SPECIALIZED MEASUREMENTS

DEDICATED PC INTERFACES

T-RAD

T-RAD-ANALOG

QUAD-4TRACK

MACH 6

APM (D)Power measurement

THZ-B series (-DZ models)

THZ-B series (-DA models)

QUAD-P series

N/A

UM-B series & THZ9D

Energy measurement

N/A

N/A

QUAD-E series

M6 series

M6 (with adaptor),
QE-B & PE-B series

OutputUSB & analog output

Analog output

USB & analog output

USB & analog output

Analog output

External trigger

Yes

Yes

Yes

Yes

N/A

Maximum repetition rate

N/A

N/A

1 kHz

200 kHz

Depends on the detector

Number of channels

11

4 (1 detector)

1

1

Detector Compatibility

142

gentec-eo.com/displays-and-pc-interfaces

ALL-IN-ONE DETECTORS
Overview of the different models

INTEGRAThe INTEGRA version of our standard laser power or energy detectors allows you toread your measurements directly on your PC thanks to our free software.

POWER DETECTORSWe also offer displays and PC interfaces which are integrated with the detector head. We offer four families of these all-in-one
detectors. INTEGRA features either a USB or RS-232 output for a direct connection to your PC. BLU is available for all our thermal power
detectors and allows you to view and log power measurements on your mobile device or PC. PRONTO includes a display, so you haveeverything you need in a single, portable device.

Simply carry your all-in-one detector and plug it in your PC any time you need tomeasure your laser power or energy. No need to buy a separate meter!
ENERGY DETECTORSU SB LASER POWER OR ENERGY METER

BLU

BEAM PROFILING

Our thermal power detectors (UP and HP series) are available in their BLU version,
which allows you to read your power measurement directly on your mobile phoneor PC thanks to Bluetooth connectivity.
You get the same high accuracy measurements without the need to connect any
wires or to carry a separate acquisition & readout device. This solution is not only
more practical, but also more economical compared to our other laser power
measurement systems.

W IRELESS LASER POWER METER

Our PRONTO series is of high interest for those who need a laser measurement
system that is portable and compact. These products can be handheld (for low
power only) or placed on a stand like our standard detectors.

P ORTABLE, ALL-IN-ONE LASER POWER METERSHP

If you prefer going wireless, the HP detectors are also available with the BLU option.

A LL-IN-ONE SOLUTIONS FOR HIGH POWER MEASUREMENTPRODUCT GUIDE 2024

T 418.651.8003 | info@gentec-eo.com.com

CUSTOM / OEM PRODUCTS

Our HP series of high power detectors include internal signal processing and two
data output options: USB to read and log measurements with your computer, or
DB15 to use a Gentec-EO display such as MAESTRO.

DISPLAYS & PC INTERFACES

These user-friendly products are so simple to use that anyone can start using themwithin seconds. They all offer data logging on their internal memory. Data can thenbe transferred to your PC via USB.

TERAHERTZ DETECTORSPRONTO143

MIRO ALTITUDE

Touchscreen, single-channel, laser power & energy meter

t

c
du

ew

o
pr

CONNECTIVITY
DetectorN

Analog out
Sync. out
External trigger

RS-232

USB-C(PC & charging)

Power supply
2x USB

Ethernet
GPIO (coming soon)

KEY FEATURESACCESSORIES	READS ALL HEADSPower: thermopiles, photodetectors and pyroelectrics
Energy: thermopiles (in SSE mode) and pyroelectrics
	LARGE TOUCHSCREEN DISPLAY
10in diagonal
1280 x 800 resolution
Touchscreen controls

Additional 12V power supply

Power cord extension

USB-C wall charger(US only)

Extra USB-C to USB-A

Pelican carrying case

STAND-R-443

	INTUITIVE USER INTERFACE
Easy to navigate interface, with 3 display modes:
scope, needle and bar chart.
Instant access to the detector settings
	REAL-TIME STATISTICAL FUNCTIONS
Max, min, average, standard deviation, RMS and PTP
stability, and repetition rate	MULTIPLE OUTPUTS
Multiple USB ports for computer connection andcharging (1x USB-C, 2x USB-A), BNC analog output,
RS-232, Ethernet, programmable I/O (coming soon)
Extra carrying sleeve

144

gentec-eo.com/displays-and-pc-interfaces

MIRO ALTITUDE
Specifications

t

uc

ew

od
pr

N

POWER DETECTORSMIRO ALTITUDE
DETECTOR TYPES

Thermopiles, pyroelectrics, photodetectors

DISPLAY10” high-resolution, anti-glare, touchscreen

POWER METER SPECIFICATIONS
4 pW to 150 kW

Meter accuracy±0.5 % ± 3 μV from 20 % to full scale

Statistics

Current value, max, min, average, standard deviation, RMS & PTP stability, time

ENERGY DETECTORSPower range

ENERGY METER SPECIFICATIONSEnergy range

2 fJ to 30 kJ

Meter accuracy

1.0 % ± 50 μV (< 500 Hz)
2.0 % ± 50 μV (500 Hz to 10 kHz)

Software trigger level

0.1 to 99.9 %, 0.1 % resolution, default 2 %Repetition rate

10 kHz for data acquisition in real time with time stamp, no missing point

Statistics

Current value, max, min, average, std dev., RMS & PTP stability, pulse #, rep. rate and average power

Thermopile

Average power & single shot energy (UP, XLP & HP series)

Photodetector

Average power (PH series)

Pyroelectric

Average power & pulse energy (QE series, except QE8)BEAM PROFILING

DETECTOR COMPATIBILITY

GENERAL SPECIFICATIONS
Analog out, 0 - 5 V (BNC)
Sync out (BNC)
RS-232 (DB9)
Ethernet (RJ45)
USB-C
2x USB-A

Rising edge external trigger

3.3-24 V (BNC)

Serial commands via

USB-C, RS-232 or Ethernet

Data storage via

Internal memory or USB key

Battery type

Rechargeable Li-ion cell

Battery life

6 hours

External power supply

12 VDC power supply included, or UBS-C (min 18 W)

DISPLAYS & PC INTERFACES

10.1-inch diagonal LCD - 1280 x 800 pixels

Outputs

TERAHERTZ DETECTORS

Digital display size

PHYSICAL CHARACTERISTICS
Mounting holes

1/4"-20 and 2x10-32 threaded holes

Dimensions

268W x 196H x 36D mm

Weight1.36 kg

ORDERING INFORMATION
Compatible stand

STAND-R-443

Product page
CUSTOM / OEM PRODUCTS

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

145

MIRO ALTITUDE
Main screen

Access the control
centerTake a screenshot

Display battery level

Start a data-logging session

Display time

1

2

1

3

NAVIGATION BAR

The upper part of the screen includes a direct access to the control center, data acquisition buttons and various indicators (battery level
and time).
2

MEASUREMENT SETTINGS PANEL

Use the various measurement settings available for your detector
to set everything related to your measurement.
Wavelength: Enter your wavelength or choose from a listof recently used wavelengths

3

DISPLAY AREA

The top part of the display area is the same for all three display
modes.
CLEAR

Range: Set the measuring range to autoscale or to one of
the standard values

Display mode: Toggle your display
mode between: scope, needle and bar
chart

Measurement mode: Choose what you want to measure:
power, SSE, moving average, etc.
Moving average: Choose the desired moving average to
use to plot the chart
Trigger: Enter the desired trigger level or choose from alist of recently used values
XNR AnticipationTM: Toggle on to measure up to 10x-20x
faster without losing any significant accuracy in your readings
Correction: Set a multiplier and an offset value for yourmeasurements
Attenuator: Toggle when using a Gentec-EO calibrated
attenuator with your detector
146

gentec-eo.com/displays-and-pc-interfaces

Clear: Use this button to reset thestatistics and erase the scope graph’s
data

ZERO

Zero: Set the current measured value
to zeroMIRO ALTITUDE
Display modes

SCOPE DISPLAY
POWER DETECTORSWith this display mode, you can travel intime using the time line at the bottom
to view measurements at any point intime while MIRO ALTITUDE continues tomeasure.
The dotted blue line shows the averagevalue.

ENERGY DETECTORSNEEDLE DISPLAY
BEAM PROFILING

Faster than an analog needle thanks to XNR
AnticipationTM! This mode is particularly
useful when tuning a laser. The real-timevalue and statistics are always visible at the
top of the screen.

TERAHERTZ DETECTORS

Arrows indicate the minimum andmaximum measured values since the lastreset. The zoom function sets these values asfull scale of the digital gauge.

BAR DISPLAYDISPLAYS & PC INTERFACES

This is the simplest display mode. Its main
advantage is that the current measured
value is displayed in huge size, allowing
you to read the measurement from a good
distance.
Arrows indicate the minimum andmaximum measured values.

CUSTOM / OEM PRODUCTS

PRODUCT GUIDE 2024

T 418.651.8003 | info@gentec-eo.com

.com

147

MIRO ALTITUDE
Settings & controls

BUILT-IN FILE MANAGER AND DATA VIEWER
MIRO’s built-in file manager lets you access and organize all your screenshots and recorded measurement sessions. You
can also copy files on your USB key.
Visualize a recorded measurement session with our built-in data viewer. Data will be displayed in the scope chart
display.
There is also a built-in image viewer so you can view your screenshots directly on your MIRO ALTITUDE.

CONTROL CENTER
The control center is accessible from all screens in the top left corner.
Easily navigate between the main screens of the app:
Display
File manager
Settings for the device.
Connect/disconnect your Gentec-EO detector to MIRO ALTITUDE

148

gentec-eo.com/displays-and-pc-interfaces

MIRO ALTITUDE
DEVICE SETTINGS
POWER DETECTORSENERGY DETECTORSBEAM PROFILING

System: Set device settings: language, date, time, sleep, number of digits, etc.
TERAHERTZ DETECTORS

Recording: Set your default recording parameters for power/energy measurement and destination.
Ethernet: Set your Ethernet parameters or let MIRO manage this automatically.
RS-232: Set your RS-232 parameters.
Analog output: Set your analog output parameters. (coming soon)

DISPLAYS & PC INTERFACES

Trigger: Use an external trigger and set your trigger level.
About: View important information about your device (serial number, firmware version, software version,calibration date) and find support.

CUSTOM / OEM PRODUCTS

PRODUCT GUIDE 2024

T 418.651.8003 | info@gentec-eo.com

.com

149

MAESTRO

Touchscreen, single channel, power & energy monitor

MULTIPLE LANGUAGESCONNECTIVITY
9VDC
USB key
Ethernet
USB (PC)

Detector

RS-232Ext. trigger
Analog out

KEY FEATURESACCESSORIESREADS ALL HEADS• Power: thermopiles, photodetectors andpyroelectrics• Energy: thermopiles (in single shot mode),
photodetectors and pyroelectrics
	LARGE TOUCHSCREEN COLOR LCDDISPLAY• 5.6in diagonal

Additional 9V power supply• FULLY touchscreen controls
UNIQUE ERGONOMIC DESIGNGreat for both handheld and tabletop use, withimproved rubber bands and kickstand for betterstability
INTUITIVE USER INTERFACE
Easy to navigate interface, with many display
features:• Single or dual graph display• Instant access to the main functions• Function search tool

• Interface available in multiple languages
USB KEY ACCESSStore data directly on a USB key
REAL-TIME STATISTICAL FUNCTIONS
Max, min, average, standard deviation, RMS andPTP stability, pulse # and repetition rate
AVAILABLE OUTPUTSUSB Key, analog output, RS-232, PC-USB,
Ethernet
150

gentec-eo.com/displays-and-pc-interfaces

Pelican carrying caseBattery pack

USB, RS-232, external trigger &analog out cables

MAESTRO
Specifications

POWER DETECTORS

MAESTRODETECTOR TYPESALL MODELS: thermopiles, pyroelectrics, photodetectors

DISPLAY

Touchscreen 5.6 in color LCD

POWER METER SPECIFICATIONSPower Range
1 µW to 30 kW

Photodetector

4 pW to 3 W

Monitor accuracy

0.25 % ± 5 µV best scale

Statistics

Current value, max, min, average, standard deviation, RMS & PTP stability, time

ENERGY DETECTORS

Thermopile

ENERGY METER SPECIFICATIONSEnergy range

2 fJ to 30 kJ± 1 % best scale

Software trigger level

0.1 to 99.9 %, 0.1 % resolution, default 2 %Repetition rate

2000 Hz / 10 000 Hz in sampling

Real-time data transfer (To USB key)

2000 HzStatistics

Current value, max, min, average, std dev., RMS & PTP stability, pulse #, rep. rate and avg power

BEAM PROFILING

Monitor accuracy

DETECTOR COMPATIBILITY
Thermopile

Average power & single shot energy

Photodetector

Average power & pulse energy

Pyroelectric

Pulse energy & average power

GENERAL SPECIFICATIONS
Digital display size

112.9 x 84.7 mm LCD - 640 x 480 pixels

Data display

Real-time, scope, statistics, digital tuning needle and averaging

Analog output

0-1 Volt, full scale, ±0.5 %Rising edge external trigger

TTL compatible, 2-25 V at 0.4 mA

Serial commands via

USB (standard), Ethernet or RS-232 (cable in option)

Data storage via

USB key

Dimensions

210W x 122H x 45D mm

Weight (with batteries)

0.67 kg

Battery type4 x rechargeable 1.2 V Ni-MH AA

Battery life

6.5 hours

External power supply

100/240 VAC 50 - 60 Hz to 9 VDC 1.66 A

DISPLAYS & PC INTERFACES

English, German, French and Japanese

TERAHERTZ DETECTORSInterface languages

ORDERING INFORMATION
Product page

CUSTOM / OEM PRODUCTS

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

151

MAESTRO
Specifications

HOME

HOME
λ

1064 nmSet Device

1064
Auto
Power
nm
OFF
-Real
3 WTime

Power

OFF

Power

ZERO
OFF

DISPLAY

Real Time

Display

Acquisition

Real TimeSet Device:

Set all the parameters related to your MAESTRO device.

Set Measure:

Set all the parameters related to your sensor.

Display:	Set the device in dual or full screen display mode and choose
the display(s) you want.

About

Acquisition:

Set all your acquisition parameters (time, sample rate, etc.).

Startup Config:

Choose how your MAESTRO will remember your sensor
settings at startup.

About:

View the main parameters and update your MAESTRO.

SET DEVICESET DEVICE

HOME
HOME

Auto - 3 W

MODE

Set Measure

Startup
Config1064 nm

RANGE

Auto - 3 W

Use the elements in this menu to set the parameters related to your MAESTRO:
Numberof Digits

SerialCommands

Ethernet

Languages

Number of Digits:

Use this menu to set the precision of the measurement.

Serial Commands:	Set compatibility with SOLO2 and use the RS-232, USB andanalog outputs

Recalibrate
Touchscreen

Ethernet:

Configure the Ethernet communication protocol.

Languages:	Select the display language: English, German, Japanese or
FrenchRecalibrate
touchscreen:

1064 nm

Auto - 3 W

Power

OFF

Real Time

Recalibrate your touchscreen by following the simple
step-by-step procedure

SET MEASURE

SET MEASURE

Use the elements in this menu to set everything related to your measurements:
Range

Wavelength

MeasureMode

Corrections

Wavelength:	Select one of the standard wavelengths offered, or entera custom value and create your own list of standard
wavelengths.
Range:

Trigger Level

Set the measuring range to autoscale or a fixed scale.

Measure Mode:	Use this menu to decide what type of measurements will be
displayed: average power, single shot energy, pulse-to-pulse
energy, etc.
Corrections:Enter multipliers and offsets.

Trigger Level:

Set the trigger level in 0.1 % steps, from 0.1 % and 99.9 %.

DUAL SCREEN DISPLAY (SHOWN WITH SCOPE DISPLAY)
1064 nm

RANGE
Auto

Power

Off

DISPLAY
Scope

19.5 sec

21.5 sec

23.5 secDATA
ACQUIS.

0.996 W
0.842 W
0.688 W
0.534 W
0.380 W
226 mW
15.5 sec

152

17.5 sec

gentec-eo.com/displays-and-pc-interfaces

25.5 sec

Any display mode can be used in both single or dual display mode. In dual display
mode, the Real-Time display takes the upper portion of the screen, while any of theother displays (Scope, Needle, Averaging or Statistics) is set on the lower portion.
The display in the lower portion can be easily changed using the parameters bar
with drop-down menus in the center of the screen. You can also expand one of thedisplays to have it in Full Screen mode using the maximize
button. Just as easily,
button.
you can go back to Dual Screen display by using the minimize

MAESTRO
Specifications

REAL-TIME DISPLAY
POWER DETECTORSThis display shows the measured value in real time, with a corresponding bar graphbelow. The large size of the digits and high contrast of the graphics allow to see themeasurement from a good distance. This mode is also always present in dual screenmode, in the upper portion of the screen.
• Very large digits• Bar graph
RANGE
Auto

1064 nm

Power

Off

DISPLAY
Real TimeDATAACQUIS.

0.996 W

With its line filling from the right of the screen, in a first-in/first-out manner, this display
mode is a good approximation of an actual oscilloscope reading. Settings include time
(x-axis) and range (y-axis). Basic statistics can also be displayed directly on the screen.

0.842 W

0.688 W

• Oscilloscope-type graph0.380 W

• On-screen, real-time statistics (min, max and average)

226 mW
15.5 sec

17.5 sec

1064 nm

RANGE
Auto

19.5 sec

21.5 sec

23.5 sec

25.5 sec

Power

Off

DISPLAYScopeDATA
ACQUIS.

BEAM PROFILING

0.534 W

• Fully customizable x and y axis

9.0

12.0

15.0

18.0

6.0

Exactly like an analog needle, only faster! This mode is particularly useful when tuning alaser. The Real-time value is also displayed at the top of the screen.

21.0
24.0

3.0

27.0

0.0

30.0

• Ultra-fast readings• Great for tuning• Real-time value at the top of the screen

Power

Off

DISPLAY
NeedleDATAACQUIS.

• Min and Max Values hold

DISPLAYS & PC INTERFACES

1064 nm

RANGE
Auto

AVERAGING DISPLAY
0.842 W

0.688 W

0.534 W

• Calculates the min, max and average values of batches of measurements

0.380 W

226 mW
15.5 sec

1064 nm115.5 sec

RANGE
Auto

PRODUCT GUIDE 2024

215.5 sec

Power

315.5 sec

415.5 secOffDISPLAY
Averaging

515.5 secDATA
ACQUIS.

• Perfect to check laser stability over time

T 418.651.8003 | info@gentec-eo.com.com

CUSTOM / OEM PRODUCTS

This very unique mode is perfect to show the trend of a laser over time. Set the numberof points per batch and let the MAESTRO identify the minimum and maximum valuesof every batch. A yellow curve then follows the average of each batch, displayed as bars
on the screen. The wider the difference between the white and blue portions of a bar(corresponding to the min and max values), the more unstable your laser is.

0.996 W

TERAHERTZ DETECTORSNEEDLE DISPLAY

Max: 25.12 mW
Min: 11.58 mW

ENERGY DETECTORSSCOPE DISPLAY153

TUNER

Single Channel, Power Monitor with Tuning Needle

DISPLAY MODE
TAIL:
Follows the speed of thepower change. The comettail is longer for fasterreading changes andshorter for slower readingchanges.
BAR GRAPH:
Fills the needle display up
to the real-time value (bestmode when viewing from a
distance).
HIGH/LOW:
When activated, indicatesthe highest and lowest
powers since activation. The
high and low needles blink
to help distinguish themfrom the real-time value.

KEY FEATURESACCESSORIESULTRA-FAST NEEDLE
Less than 1 second response time
READS ALL POWER DETECTORS
Thermopiles and photodetectors of the PH100 andPH20 Series
LARGE LCD DISPLAY• 77 x 58 mm• 17.5 mm digits• Backlight (with AC adaptor)
3 DISPLAY FUNCTIONS FOR THE NEEDLE• Normal• Tail Mode (indicates speed)• Bar graphAlso HIGH and LOW values hold
SINGLE-BUTTON NAVIGATION
Direct access and long press access to the mainfunctions
LOW CONSUMPTION
Lasts 500 hours with 4 AA alkaline batteries154

gentec-eo.com/displays-and-pc-interfaces

Additional 9V power supply

Wall support

Pelican carrying case

TUNER
Specifications

POWER DETECTORS

TUNER
DETECTOR TYPES

Thermopiles, photodetectors (PH series)

DISPLAY

LCD with tuning needle and backlight

POWER METER SPECIFICATIONSPower range

10 pW to 10 kW

Digital resolution
10 pW

XLP series

1 µW

UP series

1 mW

HP series

100 mW (HP60A), 1 W (HP100A)

Monitor accuracy± 1 %, full scale

Statistics

Min, max

Response time

<1s

ENERGY DETECTORSPH series

DETECTOR COMPATIBILITY
Average power (W, dBm)

Photodetectors (PH series)

Average power (W, dBm)

BEAM PROFILING

Thermopiles
GENERAL SPECIFICATIONS

77 x 58 mm LCD

Needle display

Ultrafast tuning needle

Needle accuracy

0.9 %Refresh rate

4 Hz

Analog output

0-1 Volt, full scale, ± 1 %Dimensions (without stand)

210W x 122H x 44D mm

Weight (with batteries)

0.47 kg

Battery type4 x AA alkaline

Battery life (estimated)

500 hours with detector

External power supply

100/240 VAC 50 - 60 Hz to 9 VDC 1.66 A

TERAHERTZ DETECTORS

Digital display size

DISPLAYS & PC INTERFACESORDERING INFORMATION
Product page

CUSTOM / OEM PRODUCTS

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

155

UNO

Single channel, power monitor

NOW AVAILABLE

W/dBm

KEY FEATURES

You can toggle your display between Watts or dBm units

ACCESSORIESREADS ALL POWER DETECTORS
Thermopiles and photodetectors of the
PH Series
LARGE LCD DISPLAY• 76 x 57 mm• 32 mm digits
UNIQUE ERGONOMIC DESIGNGreat for both handheld and tabletop use
ACCURATE
24 bit A/D converter for high resolution
measurements
SINGLE-BUTTON NAVIGATION
Direct access and long press access to all thefunctions
EXTREMELY LOW CONSUMPTION
Lasts 670 hours with 4 AA alkaline batteries
ECONOMICAL
Get the best value for your money with thisinexpensive and simple to use power monitor

156

gentec-eo.com/displays-and-pc-interfaces

Optional 9V power supply

Wall support

Pelican carrying case

UNO

Specifications

POWER DETECTORSUNO
DETECTOR TYPES

Thermopiles, photodetectors (PH series)

DISPLAY

LCD

POWER METER SPECIFICATIONSPower range

10 nW to 10 kW
Single wide range scale

Photodetector

Autoscale

ENERGY DETECTORS

Thermopile
Digital resolution
PH series

1 pW

XLP series

1 µW

UP series

1 mW

Monitor accuracy± 1%

Response time

1s

DETECTOR COMPATIBILITY
Average power (W, dBm)

Photodetectors

Average power (W, dBm)

BEAM PROFILING

Thermopiles
GENERAL SPECIFICATIONS

76 x 57 mm LCD

Digit height

32 mm

Digit typeHigh contrast fields

Data display

Real-time

Dimensions (without stand)

210W x 122H x 44D mm

Weight (with batteries)

0.47 kg

Battery type4 x AA alkaline

Battery life (estimated)

670 hours with detector

External power supply (optional)

100/240 VAC 50 - 60 Hz to 9 VDC 1.66 A

TERAHERTZ DETECTORS

Digital display size

ORDERING INFORMATION
Product pageDISPLAYS & PC INTERFACESCUSTOM / OEM PRODUCTS

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

157

U-LINK

Single channel, PC-based universal power and energy monitor

CONNECTIVITY
U-LINK (USB)

Detector

USB (PC)
External trigger
Sync. out
Analog out
U-LINK (RS-232)

Detector
Power supply
RS-232
Analog out

KEY FEATURESTHE UNIVERSAL PC-BASED METER
Reads ALL heads:

ACCESSORIES• Power: thermopiles, photodetectors andpyroelectrics• Energy: thermopiles (in single shot mode),
photodetectors and pyroelectrics
MEASURE fJ ENERGY LEVELS
Thanks to a unique digital method for suppressing
the noise on the lower ranges
EXTERNAL TRIGGER
Synchronize your U-LINK to your pulsed laser ordigital chopper (available on U-LINK (USB) model
only)
SYNCHRONIZE MULTIPLE CHANNELS
With the “SYNC. OUT” port, you can plug multiple
U-LINK devices together and create a low-cost multichannel system (available on U-LINK (USB) model
only)
SERIAL COMMANDS
Serial commands are available on both versions to let
you take full control
REAL-TIME STATISTICAL FUNCTIONS
Max, min, average, standard deviation, RMS and PTP
stability.
158

gentec-eo.com/displays-and-pc-interfaces

USB, RS-232,
external trigger & analog
out cables

Pelican carrying case

Additional 9V power supply(RS-232 version only)U-LINK
Specifications

POWER DETECTORS

U-LINK (RS-232)

DETECTOR TYPESALL MODELS: thermopiles, pyroelectrics, photodetectors

ALL MODELS: thermopiles, pyroelectrics, photodetectors

DISPLAY

1-Channel / PC-based

1-Channel / PC-based

Power range

4 pW to 30 kW

4 pW to 30 kW

Resolution (digital)

23 bits on current scale

23 bits on current scale

Monitor accuracy±0.5 % ± 3 μV

±0.5 % ± 3 μV

Statistics

Current value, max, min, average, std dev., RMS & PTP stability, time

Current value, max, min, average, std dev., RMS & PTP stability, time

Energy range

2 fJ to 30 kJ

2 fJ to 30 kJ

Resolution (digital)

Current scale/3754

Current scale/3754

Monitor accuracy

1 % ± 50 μV (< 500 Hz) / 2 % ± 50 μV (500 Hz - 10 kHz)

1 % ± 50 μV (< 500 Hz) / 2 % ± 50 μV (500 Hz - 10 kHz)

Software trigger level

0.1 to 99.9 %, 0.1 % resolution, default 2 %0.1 to 99.9 %, 0.1 % resolution, default 2 %Repetition rate

10 kHz

10 kHzReal-time data transfer

10 kHz with time stamp, no missing point

10 kHz with time stamp, no missing point

Statistics

Current value, max, min, average, std dev., RMS & PTP stability,
pulse #, repetition rate, average power

Current value, max, min, average, std dev., RMS & PTP stability,
pulse #, repetition rate, average power

Thermopile

Average power & single shot energy

Average power & single shot energy

Pyroelectric

Pulse energy & average power

Pulse energy & average power

Photodetectors

Average power & pulse energy

Average power & pulse energy

Digital display

Computer screen

Computer screen

Data display

With PC-Gentec-EO: real-time, scope, averaging, statistics and digital With PC-Gentec-EO: real-time, scope, averaging, statistics and digitaltuning needle
tuning needle

POWER METER SPECIFICATIONSENERGY METER SPECIFICATIONSa

BEAM PROFILING

DETECTOR COMPATIBILITY

USB

Real-time data transfer rate

RS-232Up to 10 kHz with time stamp, no missing point (for pyroelectrics only)

0 - 2 V, full scale, ± 1 %, user-defined0 - 2 V, full scale, ± 1 %, user-definedExternal trigger

3.3 to 12 V

3.3 to 12 V

Dimensions

57W x 26H x 91D mm

57W x 26H x 91D mm

Weight0.12 kg

0.12 kgDISPLAYS & PC INTERFACES

Up to 10 kHz with time stamp, no missing point (for pyroelectrics only)

Analog outputa

TERAHERTZ DETECTORSGENERAL SPECIFICATIONS

Serial commands and data transfer via

ENERGY DETECTORS

U-LINK (USB)

ORDERING INFORMATION
Product pagea. Maximum repetition rate and data transfer rate may vary with PC and detector speeds.
CUSTOM / OEM PRODUCTS

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

159

S-LINK

Dual & single channel, PC-based power and energy monitor

CONNECTIVITY
S-LINK-1(Ethernet)
Detector
External trigger
Power supply
USB (PC)
Ethernet
S-LINK-2(Ethernet)

Detector
Detector
External trigger
Power supply
USB (PC)
Ethernet

KEY FEATURESACCESSORIESREADS BOTH POWER AND ENERGY
Thermopiles and pyroelectrics
AVAILABLE WITH 1 OR 2 CHANNELS
S-LINK-1 and S-LINK-2 models now available
PC-BASED
Connects to your PC with included software
SERIAL COMMANDS
Serial commands are available on all versions to
let you take full control
FASTEST DATA TRANSFER RATE
Get all the points transferred directly into your PC
at 10 kHz/channel
USB OR ETHERNET
Choose your favourite communications port.
EXTERNAL TRIGGER
Every model comes standard with a 2.4 V to 24 V
external trigger

160

gentec-eo.com/displays-and-pc-interfaces

Additional 9V power supply

USB cable

Pelican carrying case

S-LINK
Specifications

POWER DETECTORSS-LINK-2

DETECTOR TYPES

Thermopiles, pyroelectrics

Thermopiles, pyroelectrics

CHANNELS / DISPLAY

1-Channel / PC-based

2-Channels / PC-based

Power range

1 µW to 10 kW

1 µW to 10 kW

Monitor accuracy±0.75 % for 10 % to full scale±0.75 % for 10 % to full scale

Statistics

Current value, max, min, average, std dev., RMS & PTP stability, time Current value, max, min, average, std dev., RMS & PTP stability, time

Response time

1s

1s

Energy range

8 fJ to 20 kJ

8 fJ to 20 kJ

Resolution (digital)

Normal mode: Current scale/4096

Normal mode: Current scale/4096< 500 Hz (MB), < 1200 Hz (MT)1%

1%500 to 1200 Hz (MB)2%2%1200 to 6000 Hz (MT)3%3%6000 to 10 000 Hz (MT)6%6%Real rime data transfera

10 kHz in normal mode, no missing point

10 kHz/Channel in normal mode, no missing point

Statistics

Current value, max, min, average, std dev., RMS & PTP stability, pulse #, repetition rate, average power

POWER METER SPECIFICATIONSENERGY METER SPECIFICATIONSMonitor accuracy

BEAM PROFILING

DETECTOR COMPATIBILITY
Thermopile

Average power & single shot energy

Average power & single shot energy

Pyroelectric

Pulse energy

Pulse energy

Number of channels

12Digital display

Computer screen

Computer screen

Data display

Real-time, ratio, line plot, histogram, statistics and 3D histogram

Real-time, ratio, line plot, histogram, statistics and 3D histogram

Serial commands and data transfer via

USB or Ethernet

USB or Ethernet

Real-time data transfer rate

10 kHz/channel in normal mode, no missing point(for pyroelectrics only) a

10 kHz/channel in normal mode, no missing point(for pyroelectrics only) a

Rising edge external trigger

3-24 V at 13 mA, optically isolated

3-24 V at 13 mA, optically isolated

Dimensions

106W x 34H x 147D mm

106W x 34H x 147D mm

Weight0.424 kg

0.424 kg

Ext. power supply

100/240 VAC 50 - 60 Hz to 9 VDC 1.66 A

100/240 VAC 50 - 60 Hz to 9 VDC 1.66 A

TERAHERTZ DETECTORSGENERAL SPECIFICATIONSDISPLAYS & PC INTERFACESORDERING INFORMATION
Product page

CUSTOM / OEM PRODUCTS

a. Actual rate may depend on the computer.

PRODUCT GUIDE 2024

ENERGY DETECTORSS-LINK-1

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

161

P-LINK

1 and 4 channels, PC-based power monitors

CONNECTIVITY
P-LINK (USB)

Detector
Analog out
USB (PC)

P-LINK (RS-232)

Detector
Analog out
Power supply
RS-232

P-LINK-4

KEY FEATURES

Detector
Detector
Detector
Detector

ACCESSORIESREADS ALL POWER DETECTORS TYPES
Thermopiles and photodetectors of the PH Series
PC-BASED
Connects to your PC with included software
MULTI-CHANNEL CAPABILITIES
Available with 1 or 4 channels
SERIAL COMMANDS
Serial commands are available on both versions to let
you take full control
REAL-TIME STATISTICAL FUNCTIONS
Max, min, average, standard deviation, RMS and PTP
stability. Also high low alarm and post-analysis mode
(P-LINK-4 only)
USB OR RS-232
Choose your favourite communications port. The USB
version is port-powered.

162

gentec-eo.com/displays-and-pc-interfaces

Additional 9V power supply(RS-232 version only)

USB & RS-232 cables

Pelican carrying case

P-LINK
Specifications

P-LINK (RS-232)

P-LINK-4 (USB)DETECTOR TYPES

Thermopiles, photodetectors

Thermopiles, photodetectors

Thermopiles, photodetectors

CHANNELS / DISPLAY

1-Channel / PC-based

1-Channel / PC-based

4-Channels / PC-based

Thermopile

3 µW to 10 kW

3 µW to 10 kW

3 µW to 30 kW

Photodetector

1 nW to 3 W

1 nW to 3 W

1 pW to 3 W

Monitor accuracy±0.5 % full scale±0.5 % full scale±0.5 % full scale

StatisticsaCurrent value, max, min, average, std dev.,
RMS & PTP stability, timeaCurrent value, max, min, average, std dev.,
RMS & PTP stability, timeb

Response time

1s

1s

1s

Thermopile

Average power & single shot energy

Average power & single shot energy

Average power

Photodetector

Average power (mW, dBm)

Average power (mW, dBm)

Average power (mW)

Number of channels

1

1

4

Digital display

Computer screen

Computer screen

Computer screen

Data displayaReal-time, histogram, statistics,
Digital tuning needleaReal-time, histogram, statistics,
Digital tuning needlebAnalog output

0 - 2 Volt, adjustable, full scale, ± 1 %0 - 2 Volt, adjustable, full scale, ± 1 %N/ASerial commands and data transfer via

USB

USB

USB

Real-time data transfer rate

10 Hz

10 Hz

10 Hz

Dimensions

57W x 26H x 91D mm

57W x 26H x 91D mm

286W x 233H x 43D mm

Weight0.12 kg

0.12 kg

2.5 kgExternal power supply

100/240 VAC 50 - 60 Hz to 12 VDC 200 mA

100/240 VAC 50 - 60 Hz to 12 VDC 200 mA

100/240 VAC 50 - 60 Hz to 5 VDC, 3 A

POWER DETECTORS

P-LINK (USB)

POWER METER SPECIFICATIONSPower range

DETECTOR COMPATIBILITY

Real-time, graphic, statistics, high/low alarm,
Post-analysis mode, multi-channel

BEAM PROFILING

GENERAL SPECIFICATIONSENERGY DETECTORS

Current value, max, min, average, std dev.,
RMS & PTP stability, time

TERAHERTZ DETECTORSORDERING INFORMATION
Product pageDISPLAYS & PC INTERFACES

a. Using PC-Gentec-EO software.
b. Using Octolink software.

CUSTOM / OEM PRODUCTS

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

163

M-LINK

Single channel, PC-based universal power and energy monitor

CONNECTIVITY
M-LINK

Analog output

USB

External trigger

KEY FEATURESACCESSORIESTHE UNIVERSAL PC-BASED METER
Reads ALL heads:• Power: thermopiles, photodetectors and pyroelectrics• Energy: thermopiles (in single shot mode),
photodetectors and pyroelectrics

MEASURE fJ ENERGY LEVELS
Thanks to a unique digital method for suppressing thenoise on the lower ranges
EXTERNAL TRIGGER
Synchronize your M-LINK to your pulsed laser
or digital chopper
DIGITAL (USB) OUTPUT
Connect the M-LINK module directly to your PC
POWERFUL LABVIEW SOFTWARE
Features include:• Complete instrument controls: range, trigger,
wavelength, etc.
• Live display in J and J/cm2 or W and W/cm2• Full Statistics: min, max, mean, standard deviation,
RMS stability, repetition rate, etc.
• Graphic displays: strip chart, histogram, tuning needle
and more• Data file collection and analysis
164

gentec-eo.com/displays-and-pc-interfaces

USB cable

Pelican carrying case

M-LINK
Specifications

POWER DETECTORS

M-LINK
DETECTOR TYPESALL MODELS: thermopiles, pyroelectrics, photodetectors

DISPLAY

PC-based

POWER METER SPECIFICATIONS
4 pW to 30 kW

Resolution (digital)

Current scale/3000

Monitor accuracy±0.5 % ± 2 digits

Statistics

Current value, max, min, average, std dev., RMS & PTP stability, time

ENERGY DETECTORSPower range

ENERGY METER SPECIFICATIONSEnergy range

30 fJ to 30 kJ

Resolution (digital)

Current scale/3000

Monitor accuracy

1 % ± 2 digits (< 1 kHz)

Software trigger level

0.1 to 99.9 %, 0.1 % resolution, default 2 %Repetition rate

1000 Hza1000 Hz with time stamp, no missing point

Statistics

Current value, max, min, average, std dev., RMS & PTP stability, pulse #, repetition rate, average power

BEAM PROFILING

Real-time data transfer
DETECTOR COMPATIBILITY
Thermopile

Average power & single shot energy

Pyroelectric

Pulse energy & average power

Photodetectors

Average power & pulse energy

GENERAL SPECIFICATIONS
Real-time, scope, averaging, statistics and digital tuning needle

Serial commands and data transfer via

USB

Real-time data transfer rate

1000 Hz with time stamp, no missing point (for pyroelectrics only)

Analog output

0 - 2 V, full scale, ± 2 % (joulemeters) ± 4 % (wattmeters)

Rising or falling edge external trigger

4.5 to 10 V @ 20 mA, optically isolated

Dimensions

106W x 34H x 147D mm

Weight0.424 kgDISPLAYS & PC INTERFACES

Computer screen

Data display

TERAHERTZ DETECTORS

Digital display

ORDERING INFORMATION
Product pagea. Maximum repetition rate may vary with PC and detector speeds.

CUSTOM / OEM PRODUCTS

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

165

INTEGRA
Embedded PC interface

CONNECTIVITY
Three models available: 				USB output (-INT)
RS-232 output (-IDR)
USB with external trigger (-INE)
USB model name -INT
To detector
To PC

RS-232 model name -IDR

Power supply

To detector

To PC oracquisition system

External trigger model name -INE

External
trigger BNC

To detector

KEY FEATURES

To PC

ALL-IN-ONE DETECTOR + METERPlug your detectors directly into your PC with the
INTEGRA embedded PC interface

ACCESSORIESINCREDIBLE PERFORMANCEINTEGRA detectors offer the same performanceas the usual detector + PC interface combination
USB OR RS-232
INTEGRA detectors are offered with a choice of
USB or RS-232 connector
COMPACT SIZE
Perfect for the lab, OEM applications and field
servicing. No need to carry a meter!

Adaptateur USB-A à USB-C

Convertisseur RS-232 à USB-A

EASY TO MOUNTLOWER RECALIBRATION COSTS
One product = one calibration. Reduce yourrecalibration costs by half!
UNIVERSAL SOFTWARE-PC-GENTEC-EO
Control your INTEGRA detector with the same
powerful software as the MAESTROCUSTOMIZABLE
Contact us for custom cable lengths andserial commands

Secure it on your optical table

WATCH OUT FOR THIS LOGO!
Available with

166

gentec-eo.com/displays-and-pc-interfaces

PC-GENTEC-EO

Universal software for INTEGRA, MAESTRO, P-LINK, U-LINK AND HP

MAIN CONTROLS
Complete and easily navigable software interface with all the necessary options and tools:
Connect or Disconnect your device.

Controls:	Turn the Turbo Mode ON or OFF, make a Zero to remove the thermal offset, start
the Acquisition of the data and start the calculations of the Statistics associatedwith this data.
Startup Config:	Save your measurements settings or Load the settings associated with an alreadyexisting file.
Get information about the PC-Gentec-EO software and read the user manual.

Measure:Configure the parameters related to your measurements.

Display: 	Set the desired number of digits and settings associated with the selected display.
Acquisition:

Enter the parameters related to the acquisition of data.

MEASUREMENT PARAMETERS

ENERGY DETECTORSHelp:

POWER DETECTORS

Connection:

The Measure tab allows you to configure the parameters related to your measurements:

Range:	Set the power or energy Range to a fixed scale or let the software
automatically adjust the scale.
Measure Mode:	Select the type of Measurement that will be displayed (power, energy) and let
the software know if you want Anticipation and if there is any Attenuation.
Corrections:

Apply a Multiplication Factor and/or an Offset to your measurements.

Trigger Level:

Set the Trigger Level in 0.1 % steps, from 0.1 % to 99.9 % (in energy mode only).

Select the display that suits you best and watch your measurements in real time! With the options
toolbar in the bottom of the interface, you can manage the displays at your convenience:
Real Time

Histogram

NeedleStatistics

Real-time:

Real-time value and corresponding bar graph

Scope:

Line filling graph

Needle:Fast analog-like needle

Averaging:Shows trend of laser over time

Histogram:

Displays up to 100 bars

Statistics:

Min, Max, Average, RMS and PTP Stability, Rep. Rate and Standard Deviation

DISPLAYS & PC INTERFACES

Averaging

Scope

DATA ACQUISITION
The Acquisition tab allows you to enter the parameters for data acquisition:

Energy Mode:	Choose a sampling rate (1 pulse out of X pulses), a total duration (total numberof pulses) for the acquisition of data, a timestamp for each value, the file nameand file location. Decide if you want to save raw data and/or the statistics
associated with this data.
T 418.651.8003 | info@gentec-eo.com.com

CUSTOM / OEM PRODUCTS

Power Mode:	Choose a Sampling Rate (number of measurements per interval of time), a
Total Duration (in days, hours, minutes and seconds) for the data acquisition,a Time Stamp for each value and the File Name and File Location. You canchoose to save only the raw data and/or the statistics associated with your dataacquisition.

PRODUCT GUIDE 2024

TERAHERTZ DETECTORSMULTIPLE DISPLAYSBEAM PROFILING

Wavelength:	Enter the Wavelength of your laser and the software will apply theappropriate correction factor on the measurements.

167

BLU

Wireless Bluetooth® PC interface

MEASURE WITH YOUR SMARTPHONE, TABLET OR PC

BLU
wireless module

Bluetooth Low
Energy (BLE)
Communication

Display the results on your mobile device with the Gentec-EO BLU
app available FREE on Google Play and Apple Store. Need to use itwith a PC? Simply plug in the included Bluetooth receptor and use
PC-Gentec-EO.

Power button
to extend
battery life
UP power detector
Recharge via USB

The Bluetooth® word mark and logos are registered trademarks owned by the Bluetooth SIG, Inc. andany use of such marks by Gentec-EO is under license.

KEY FEATURES
ALL-IN-ONE DETECTOR + METER
This new line of All-in-One detectors combine a
detector and a meter with Bluetooth connectivity in one
convenient product. No need to carry a meter!
SAVE 50 % ON CALIBRATION COSTS
One product = one calibration. Reduce your recalibration
costs by half!
EXTENSIVE COVERAGE
Receive data at up to 30m from the detector, with thesame performance as the usual detector + PC interface
combination.
EASY TO SET UP
Perfect for field service, labs and OEM applications.
GO WIRELESSNo need to worry about cable length or PC interface
location.
LONG BATTERY LIFEThe USB-rechargeable Li-ion battery lasts up to
5 continuous days with the device running
168

gentec-eo.com/displays-and-pc-interfaces

WATCH OUT FOR THIS LOGO!
Available with

BLU

Free software for smartphones and tablets

MAIN CONTROLS

If there are no devices within range, the app will propose a simulator.
When a BLU detector is connected to a phone or computer, no other device cancommunicate with it.

The menu tab, available with theicon or by swiping from the left of the screen, allows
you to configure the parameters related to your measurements:
Wavelength:	Enter the wavelength of your laser and the software will apply theappropriate correction factor on the measurements.

ENERGY DETECTORSMEASUREMENT PARAMETERS

POWER DETECTORS

Connecting a BLU device is very simple in the mobile application. Just open the app
and it will automatically search for all available devices. Then, tap on the desired
device in the list.

Range:	Set the power or energy range to a fixed scale or let the software
automatically adjust the scale.

Corrections:	Apply a multiplication factor and/or an offset to your measurements.
Trigger Level:	Set the trigger level in 0.1 % steps, from 0.1 % to 99.9 % (in energy
mode only).
Connection:	Use this option to see the list of BLU devices within range.

Select the display that suits you best and watch your measurements in real time! Simply
swipe the screen to switch between the various displays:
Scope:	Line filling graph; grab screenshots to save & share easily with yourdeviceDISPLAYS & PC INTERFACES

Needle:	Fast analog-like needle
Real-time:	Real-time value and corresponding bar graphStatistics:	Min, max, average, RMS and PTP stability and standard deviation

DATA ACQUISITION
Power mode:	Choose a sample rate (number of measurements per interval of time) anda duration (in days, hours, minutes and seconds) for the data acquisition.

Energy mode:	Choose a sampling rate (1 pulse out of X pulses) and a duration (total
number of pulses) for the acquisition of data.

Once the acquisition is complete, you can export the data to any of the data-sharing apps
installed on your mobile device, or send it by email

T 418.651.8003 | info@gentec-eo.com.com

CUSTOM / OEM PRODUCTS

The acquisition screen allows you to enter the parameters for data acquisition:

PRODUCT GUIDE 2024

TERAHERTZ DETECTORSMULTIPLE DISPLAYSBEAM PROFILING

Measure Mode:	Select the type of measurement that will be displayed: power, in watts(default) or single shot energy, in joules (energy/calorimeter mode).

169

CUSTOM / OEM PRODUCTS
Product characteristics

Gentec-EO offers OEM customers the highest flexibility so that you make no compromise. Whether you want a different housing, aspecific sensitivity or another output connector, we have a solution for you. We will customize existing models or design a whole newdetector to meet your needs.

COMPACTNESSAs an OEM, we know space is often a constraint. This is why we offer very compact
detectors to ease the integration inside machines. Users can mix and matchexisting detectors and cooling modules from a large set of combinations.

Ø 12

38

34.1

(4) N/A(3) Ground/Shield(2) Supply 12 VDC(1) Signal Out

38

PERFORMANCE
Anticipation

Amplification

Filtering

0-95 % of the signal in as
quickly as 0.3 s withthe small UD12-70-H5
and in 0.6 s with theUD19-150-H5 using our
external PCB.

Adjust your disk sensitivity
to get the perfect voltage
for your acquisition
system. Disks can be
adjusted from 0.01 V/W
up to 10 V/W dependingon the model.

Eliminate the highfrequency noise comingfrom the environment
with the integrated lowpass filter of our PCB.

CONNECTIVITY
Gentec-EO offers you several types of output connectors, from the more standard
DB15, BNC and Molex to any exotic type you may need.

170

gentec-eo.com/customization

DB15

BNC

Molex or bare wires

This connector contains
an EEPROM with custom
calibration data forboth power and energy
detectors.

The BNC output gives you
fast, easy installation anddirect connection to an
oscilloscope

The internal PCB gives anamplified signal output
that can be accessed via
a Molex connector andcable or bare wires. It isconvenient for integrated
systems.

CUSTOM / OEM PRODUCTS
Overview of the different models

Almost anything you see in our product line can be turned into an OEM unit! We also offer standard OEM products, at different levels of
integration: from the simple thermopile disk to a complete head with internal PCB for signal anticipation and amplification.
POWER DETECTORSUD SERIES• Thermal sensor disks• Designed for integration• Many sizes and absorber choices:

ENERGY DETECTORS

10, 12, 19, 25, and 55 mm Ø apertures
Broadband or high damage threshold coatings

T HERMAL SENSOR DISKS

Available with

BEAM PROFILING

UP SERIES• Complete thermal heads with cooling modules• Several sizes, coolings and absorber choices:

TERAHERTZ DETECTORS

Apertures from 10 to 55 mm Ø
Broadband or high damage threshold coatings
Convection, fan or water-cooled• DB15, USB, RS-232 or wirelessT HERMAL SENSOR HEADSDISPLAYS & PC INTERFACESUP SERIES WITH PCB• Complete thermal heads with cooling modules• Internal PCB for amplification, anticipation and filtering• Several sizes, coolings and absorber choices:

CUSTOM / OEM PRODUCTS

10, 12, 19, 25, 50 and 55 mm Ø apertures
Broadband or high damage threshold coatings
Convection, fan or water-cooled• DB15, BNC or Molex connector, or bare wires

T HERMAL SENSOR HEADS WITH PCBPRODUCT GUIDE 2024

T 418.651.8003 | info@gentec-eo.com

.com

171

UD

Thermal sensor disks, 10 - 55 mm Ø, 200 μW - 500 W

HOW TO USE SENSOR DISKSThe UD thermal sensor disks were designed for integration into laser
systems. They are the solution if you are engineering the cooling andsignal processing into your system already.
The chart below and on the next page show the various possibilities
that Gentec-EO offers to OEM users. The choice of a level of integration
depends on your needs in terms of calibration, output signal level,
cooling avaibility, etc.

Disk Alone• Thermal sensor disk

Disk + PCB• Thermal sensor disk
• Amplification - anticipation - filtering

KEY FEATURESDESIGNED FOR INTEGRATIONWith a broad bandwidth and high power
densities
VERY THIN PROFILES
Starting at only 2 mm deep
VARIOUS APERTURE SIZES
Choose your aperture from 10 mm Ø to 55 mm Ø
2 LEVELS OF INTEGRATION• Disk alone• Disk + PCB
CUSTOM PRODUCTS
Contact us for more options!

172

gentec-eo.com/customization

UD

Specifications

UD12-70-H5

UD19-50-W5

UD19-200-H9

UD25-350-H12

UD55-700-HD

2W/2W

70 W / 30 W

50 W / 50 W

200 W / 110 W

350 W / 250 W

700 W / 600 W

10 mm Ø12 mm Ø19 mm Ø19 mm Ø25 mm Ø55 mm ØSpectral range

0.19 - 20 μm

0.19 - 20 μm

0.19 - 10 μm

0.19 - 20 μm

0.19 - 20 μm

0.19 - 20 μm

Noise equivalent power

0.1 mW

1 mW

1 mW

3 mW

10 mW

45 mW

Rise time (nominal) a, b

3.0 s

1.6 s

5s

4.5 s

7.9 s

14 s

2 mV/W

0.53 mV/W

0.65 mV/W

0.23 mV/W

0.1 mV/W

0.03 mV/W

2.4 mV/J

0.84 mV/J

0.33 mV/J

0.23 mV/J

0.05 mV/J

0.008 mV/J

Maximum measurable energy 3 J

5J

200 J

25 J

40 J

200 J

Noise equivalent energy a

20 mJ

23 mJ

60 mJ

200 mJ

250 mJ

36 kW/cm2

100 kW/cm2

45 kW/cm2

45 kW/cm2

45 kW/cm2

MAX AVERAGE POWER(WATER-COOLED / FAN-COOLED)

EFFECTIVE APERTURE

POWER DETECTORS

UD10-2-H5-L

MEASUREMENT CAPABILITY

ENERGY DETECTORS

Sensitivity (typ into 100 kΩ load)bEnergy mode
Sensitivityc

5 mJ

DAMAGE THRESHOLDSBEAM PROFILING

Maximum average power density 36 kW/cm2
Maximum energy density
1064 nm, 360 µs, 5 Hz

5 J/cm2

5 J/cm2

100 J/cm2

9 J/cm2

9 J/cm2

9 J/cm2

1064 nm, 7 ns, 10 Hz

1 J/cm

1 J/cm

1.1 J/cm

1 J/cm

1 J/cm

1 J/cm2

532 nm, 7 ns, 10 Hz

0.6 J/cm2

0.6 J/cm2

1.1 J/cm2

266 nm, 7 ns, 10 Hz

0.3 J/cm

0.3 J/cm

Absorber

H5

Dimensions

44Ø x 3D mm

Weight (head only)

7g

2

2

2

2

2

2

0.6 J/cm2

0.6 J/cm2

0.6 J/cm2

0.7 J/cm

0.3 J/cm

0.3 J/cm

0.3 J/cm2

H5

W5

H9

H12

HD

36Ø x 2D mm

44Ø x 3D mm

44Ø x 3D mm

54Ø x 3D mm

85Ø x 4D mm

4g

7g

7g

13 g

180 g

2

2

2

2

PHYSICAL CHARACTERISTICSTERAHERTZ DETECTORSORDERING INFORMATION
Product pageDISPLAYS & PC INTERFACES

a. These characteristics depend on the thermal management and electronics provided by the user. Packaging, cooling and electronics similar to our UP Series detectors will provide similar
performances. See UP Series specifications sheets for more details. Actual performance depends on the tradeoffs in a user’s design. It may be possible to enhance some performance parametersat the expense of others.
b. Without anticipation algorithm or circuitry.
c. For 360 μs pulses. Higher pulse energy possible for long pulses (ms), less for short pulses (ns).

CUSTOM / OEM PRODUCTS

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

173

DISCRETE PYROSSpecifications

OUTPUT OPTIONSCHOOSE YOUR CONFIGURATION• QS-L: Passive discrete pyroelectric detectors with thermally isolated
crystal for high sensitivity (low noise) at low frequencies• QS-H: Passive discrete pyroelectric detectors with heat sink for highaverage power and high frequency operation• QS-IL: Current-mode hybrid sensors designed for high sensitivity, low
bandwidth applications• QS-IF: Current-mode hybrid sensors designed for high frequency
applications, up to 20 MHz

SEVERAL IR WINDOWS IN OPTION• Quartz:• Barium fluoride:• Sapphire:• Silicon:• AR germanium:

KEY FEATURES

0.2 - 3.0 μm
0.2 - 17.5 μm
0.1 - 7.0 μm
1.1 - 9.0 μm and 50 - 1000 μm
8 - 14 μm

ACCESSORIESBROAD SPECTRAL RESPONSE
From 0.1 to 1000 µm
EASY TO INTEGRATE FORMAT
TO5 and TO8 packages make the QS detectors
small and easy to integrate in an existing
systemLARGE AREA SENSORS
5 mm Ø and 9 mm Ø diameter pyroelectric
sensors make optical alignment easier.
TEST BOX AVAILABLE
Can be used with our QS-I-TEST test box which
provides mounting and power supply

QS-I-TEST
Evaluation test box (current)

Permanent IR windows(Various types available)

Pelican carrying case

QS-I-TEST EVALUATION TEST BOX
QS-I-TEST

ROOM-TEMPERATURE OPERATION

Batteries+9 V / -9 V
105 - 1010 Ω

FAST RESPONSE

Rf resistors
Cf compensating

Yes

Package

101.6H x 127L x 58.4P

Optical mount¼-20 threaded

Front bezel

SM1 (1.035-40)

Product number

201693* For details, contact your Gentec-EO representative

174

gentec-eo.com/customization

QS-L & QS-H
Specifications

POWER DETECTORS

Discrete pyro detectors, low noise level

QS2-L

QS3-L

QS5-L

QS9-L

CURRENT RESPONSIVITY

0.5 µA/W

0.5 µA/W

0.25 µA/W

0.25 µA/W

EFFECTIVE APERTURE

2 mm Ø

3 mm Ø

5 mm Ø

9 mm Ø

PACKAGE

TO5

TO5

TO5

TO8

Spectral range

0.1 - 1000 µm

0.1 - 1000 µm

0.1 - 1000 µm

0.1 - 1000 µm

Max average power

50 mW

50 mW

50 mW

50 mW

Capacitance (at 1000 Hz)

22 pF

60 pF

90 pF

250 pF

Current responsivity (at 630 nm)

0.5 µA/W

0.5 µA/W

0.25 µA/W

0.25 µA/W

Thermal frequency (3 dB)

1.6 Hz0.8 Hz0.5 Hz0.25 Hz

Temperature coefficient

0.2 %/°C

0.2 %/°C

0.2 %/°C

0.2 %/°C

MEASUREMENT CAPABILITY
ENERGY DETECTORSORDERING INFORMATION
Product page

BEAM PROFILING

Discrete pyro detectors, high average power

QS3-H

QS5-H

QS9-H

500 mW

500 mW

500 mW

500 mW

EFFECTIVE APERTURE

2 mm Ø

3 mm Ø

5 mm Ø

9 mm Ø

PACKAGE

TO5

TO5

TO5

TO8

Spectral range

0.1 - 1000 µm

0.1 - 1000 µm

0.1 - 1000 µm

0.1 - 1000 µm

Max average power

500 mW

500 mW

500 mW

500 mW

Capacitance (at 1000 Hz)

12 pF

30 pF

90 pF

250 pF

Current responsivity (at 630 nm)

0.25 µA/W

0.25 µA/W

0.25 µA/W

0.25 µA/W

Thermal frequency (3 dB)

5 Hz

5 Hz

5 Hz

5 Hz

Temperature coefficient

0.2 %/°C

0.2 %/°C

0.2 %/°C

0.2 %/°C

Effective aperture

2 mm Ø

3 mm Ø

5 mm Ø

9 mm Ø

Package

TO5

TO5

TO5

TO8

SensorPyroelectric

Pyroelectric

Pyroelectric

Pyroelectric

Absorber

MT

MT

MT

MT

Dimensions (excluding pins)

9.1Ø x 6.4D mm

9.1Ø x 6.4D mm

9.1Ø x 6.4D mm

15.2Ø x 6.4D mm

Weight

1.0 g

1.0 g

1.0 g

1.5 g

TERAHERTZ DETECTORS

QS2-H
MAX AVERAGE POWER

MEASUREMENT CAPABILITYDISPLAYS & PC INTERFACESPHYSICAL CHARACTERISTICSCUSTOM / OEM PRODUCTS

ORDERING INFORMATION
Product page

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

175

QS-IF
Specifications

Hybrid pyro detectors, current mode, fast response

QS2-IF

QS3-IF

QS5-IF

QS9-IFVOLTAGE RESPONSIVITY

50 V/W

50 V/W

25 V/W

25 V/W

CURRENT RESPONSIVITY

0.5 µA/W

0.5 µA/W

0.25 µA/W

0.25 µA/W

EFFECTIVE APERTURE

2 mm Ø

3 mm Ø

5 mm Ø

9 mm Ø

PACKAGE

TO5

TO5

TO5

TO8

Spectral Range

0.1 - 1000 µm

0.1 - 1000 µm

0.1 - 1000 µm

0.1 - 1000 µm

Max average power

50 mW

50 mW

50 mW

50 mW

Noise equivalent power a

8x10-8 W/(Hz)½8x10-8 W/(Hz)½

1.6x10-7 W/(Hz)½

1.6x10-7 W/(Hz)½

Detectivity a

2.2x106 cm(Hz)½ /W

3.3x106 cm(Hz)½ /W

2.8x106 cm(Hz)½ /W

5.0x106 cm(Hz)½ /W

Capacitance (at 1000 Hz)

22 pF

60 pF

90 pF

250 pF

Current responsivity (at 630 nm)

0.5 µA/W

0.5 µA/W

0.25 µA/W

0.25 µA/W

Voltage responsivity

MEASUREMENT CAPABILITY

50 V/W

50 V/W

25 V/W

25 V/W

Thermal frequency (3 dB)

1.6 Hz0.8 Hz0.5 Hz0.25 Hz

Feedback resistor

100 MΩ

100 MΩ

100 MΩ

100 MΩ

Supply voltage± 12 V± 12 V± 12 V± 12 V

Effective aperture

2 mm Ø

3 mm Ø

5 mm Ø

9 mm Ø

Package

TO5

TO5

TO5

TO8

SensorPyroelectric

Pyroelectric

Pyroelectric

Pyroelectric

Absorber

MT

MT

MT

MT

Dimensions

9.1Ø x 6.4D mm

9.1Ø x 6.4D mm

9.1Ø x 6.4D mm

15.2Ø x 6.4D mm

Weight

1.0 g

1.0 g

1.0 g

1.5 g

b

PHYSICAL CHARACTERISTICS

ORDERING INFORMATION
Product pagea. 630 nm, 15 Hz, 1 Hz bandwidthb. 630 nm, 15 Hz

176

gentec-eo.com/customization

QS-ILSpecifications

POWER DETECTORS

Hybrid pyro detectors, current mode, low noise level

QS2-IL

QS3-IL

QS5-IL

QS9-ILVOLTAGE RESPONSIVITY

25 kV/W

25 kV/W

13 kV/W

13 kV/W

CURRENT RESPONSIVITY

0.5 µA/W

0.5 µA/W

0.25 µA/W

0.25 µA/W

EFFECTIVE APERTURE

2 mm Ø

3 mm Ø

5 mm Ø

9 mm Ø

PACKAGE

TO5

TO5

TO5

TO8

Spectral range

0.1 - 1000 µm

0.1 - 1000 µm

0.1 - 1000 µm

0.1 - 1000 µm

Max average power

50 mW

50 mW

50 mW

50 mW

Noise equivalent power a

2x10-9 W/(Hz)½2x10-9 W/(Hz)½

6x10-9 W/(Hz)½

6x10-9 W/(Hz)½

Detectivity a

9.0x107 cm(Hz)½ /W

1.3x108 cm(Hz)½ /W

7.0x107 cm(Hz)½ /W

1.3x108 cm(Hz)½ /W

Capacitance (at 1000 Hz)

22 pF

60 pF

90 pF

250 pF

Current responsivity (at 630 nm)

0.5 µA/W

0.5 µA/W

0.25 µA/W

0.25 µA/W

Voltage responsivity

MEASUREMENT CAPABILITY

13 kV/W

13 kV/W

0.8 Hz0.5 Hz0.25 Hz

Feedback resistor

100 GΩ

100 GΩ

100 GΩ

100 GΩ

Supply voltage± 5 to ± 12 V

± 5 to ± 12 V

± 5 to ± 12 V

± 5 to ± 12 V

Effective aperture

2 mm Ø

3 mm Ø

5 mm Ø

9 mm Ø

Package

TO5

TO5

TO5

TO8

SensorPyroelectric

Pyroelectric

Pyroelectric

Pyroelectric

Absorber

MT

MT

MT

MT

Dimensions

9.1Ø x 6.4D mm

9.1Ø x 6.4D mm

9.1Ø x 6.4D mm

15.2Ø x 6.4D mm

Weight

1.0 g

1.0 g

1.0 g

1.5 g

BEAM PROFILING

25 kV/W

1.6 HzENERGY DETECTORS

25 kV/W

Thermal fequency (3 dB)b

PHYSICAL CHARACTERISTICSTERAHERTZ DETECTORSORDERING INFORMATION
Product pageDISPLAYS & PC INTERFACES

a. 630 nm, 5 Hz, 1 Hz bandwidthb. 630 nm, 15 Hz

CUSTOM / OEM PRODUCTS

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

177

DISCRETE PYROSSpecifications

PYROELECTRIC THERMAL DETECTORS
Our pyroelectric detectors are a class of room temperature thermal detectorsthat produce a current output that is directly proportional to the rate ofchange of temperature when exposed to a source of radiation. They are
best described by an AC current source, capacitor and resistor. Their current
output is governed by the equation I = p(T)·A·dT/dt, where I is current, p(T)is the pyro coefficient, A is the area as defined by the front electrode, anddT/dt is the rate of temperature change of the pyro crystal. The advantagesof a pyroelectric detector over other IR detectors are: room temperature
operation, broad spectral response, high sensitivity (D*) and fast response(sub-ns into 50 Ω).

2

2

8
1 CASE8
1 CASE

QS-L AND QS-H DISCRETE PYROS
Our passive discrete pyroelectric detectors range from 1 to 9 mm in diameter andare provided in two configurations: high sensitivity or high average power. They
present a pyroelectric detector element covered with our metallic coating (MT) andare packaged in a miniature TO-5 or TO-8 can. The diagram shown left identifies
the pin-out for both types of detectors. Our organic black coating (BL), increases the
optical absorption and helps flatten the spectral response. We also offer a number ofpermanent IR Windows that can be added to the TO can. These discrete pyro detectors
are ideal for pulsed laser applications.

QS-L (left) and QS-H (right) pin-outs

QS-IF AND QS-IL CURRENT MODE HYBRID PYROS
7

Rf
2

6

8

4

These detectors offer high gain (>105 V/W) and/or high bandwidth (>10 MHz). In
this configuration, the pyroelectric detector element is combined to a low noise
operational amplifier. The QS-IL models are designed for high performance at low to
medium frequencies, while the QS-IF models offer good performance at medium tohigh frequencies. These detectors are very easy to use. Simply supply the +/- 10 to 15 V
to power the operational amplifier and add an external resistor, if required, to adjust
the bandwidth and you are ready to measure pulsed, modulated or chopped sources,from nJ to mJ and nW to W. These detectors also make great candidates for any
variety of broadband analytical instruments or laser measurement products.

QS-IF and QS-IL pin-out

0.1 µF

REXT
CEXT

Rf

2

QS
DETECTOR

6
VOUT
8

4
0.1 µF-12V

QS-IL circuitry

178

+V

7

gentec-eo.com/customization

VOLTAGE OUTPUT VS. FREQUENCYOur QS-IL hybrid detectors are designed to maximize voltage output at low
frequencies and therefore include load and feedback resistors in the 100 GΩ to
300 GΩ range. They are also designed into 8-pin TO packages that allow the additionof an “external resistor” to lower the output and increase the bandwidth. The circuit
diagram at the left shows a typical hook up for our QS5-IL detector (with our MT
coating), using external resistors and capacitors. Our QS-IF series, on the other hand, aredesigned for high bandwidth applications and therefore include a smaller feedback
resistor of 100 MΩ. For expert help on designing a detector circuit please contact
us info@gentec-eo.com.

DISCRETE PYROSSpecifications

OPERATION IN POWER MEASUREMENT MODE
POWER DETECTORS

When using our QS-IL hybrid detector to measure the power (in W) of your CW or highrepetition rate source (quasi-CW), you will need to employ an optical chopper. The
diagram at the left shows the typical voltage output of a QS5-IL when used with our
QS-I-TEST evaluation test box. Note that the voltage output is an approximate “squarewave” whose rise and fall times are governed by the RC time constant of the circuit.
The optical power is directly proportional to the peak voltage minus the baseline
voltage. We calibrate these devices when operating in this mode.

ENERGY DETECTORS

Typical QS-IL voltage output inpower measurement mMode

OPERATION IN ENERGY MEASUREMENT MODE

BEAM PROFILING

Our pyroelectric detectors are an ideal choice when measuring the performance ofyour pulsed laser in the range of nJ to mJ, across the full spectrum! The scope trace
at the left represents the typical output from a QS9-IL, when used with our QS-I-TEST
set up as an integrating joulemeter. Note the fast rise to a peak and then slower
decay governed by the RC time constant selected for the integrating circuit. In thisconfiguration you can measure absolute pulse energy, rep rate, and pulse-to-pulse
stability. The maximum pulse width of your source is determined by the RC time
constant you select and there is no limit as to how short the pulse can be!
Typical QS-IL voltage output inenergy measurement mode

90%

BL

80%

Absorptance

70%
60%
50%

MT

40%
30%
20%
10%
0.1

1
Wavelength (µm)

10Absorption curves of
QS pyroelectric detectors

DISPLAYS & PC INTERFACES

0%

Unlike photoconductive and photovoltaic detectors, our pyroelectric thermal
detectors are not limited to a small part of the electromagnetic spectrum. They are
truly broad spectrum detectors, sensitive from 0.1 µm to 3000 µm (EUV, FAR IR, andTHz). Any and all radiation absorbed by our coatings or pyro crystal will result in ameasurable signal. The two plots at the left show the relative spectral response of
detectors with MT and BL coatings. Note that the well documented, NIST traceablecalibrated portion of these curves runs from 0.25 µm to 15 µm. There are currently no
traceable optical standards for measurements > 15 µm.

TERAHERTZ DETECTORSBROAD SPECTRAL RESPONSE

100%

CUSTOM / OEM PRODUCTS

PRODUCT GUIDE 2024

T 418.651.8003 | info@gentec-eo.com

.com

179

CUSTOM DESIGN EXAMPLES
Specifications

After over 50 years of experience in the laser beam measurementbusiness, we have developped many customized solutions, sometimes
for very unusual applications! This section is only a small portion of
the projects we have accomplished for our customers, so do nothesitate to contact us with any special need you may have. We arealways striving to find the perfect solution for your application!

180

gentec-eo.com/customization

CUSTOM DESIGN EXAMPLES
Specifications

EXTREMELY HIGH POWER, LOW BACK-REFLECTIONSThis custom project example can measure up to 100 kW of average power continuously,
and less than 4 % of the incident radiation is backscattered.

ENERGY DETECTORSCUSTOM-DESIGNED HIGH-POWER DETECTOR

POWER DETECTORS

When working at extremely high average power, even a low % of back-reflections
can be dangerous. To manage the back-reflections and provide a safer working
environment, we can equip your high-power detector with a water-cooled “TUBE”
extension.

"10 PW PORTABLE BLACK HOLE"

By working closely with our client, we have designed the only existing beam dumpthat can fulfill the task of capturing and dissipating the energy contained in the single
pulses of the ELI-NP end-of-line laser beams. Furthermore, this product was designed
to be operated without external cooling, which simplifies its installation and makes it
usable in a wide range of applications.

TERAHERTZ DETECTORSUNRIVALED DAMAGE THRESHOLDS: UP TO 200 J/CM2 FOR fs PULSESEXTREMELY LOW BACK-REFLECTIONS: < 0.02 %DISPLAYS & PC INTERFACES

8-CHANNEL ARRAY FOR THZ TOMOGRAPHY
This eight-element, pyroelectric detector array was designed for use in a THz
tomography system. It was used to analyze high pressure flames by measuring
absorption in the water spectral bands, using THz radiation in the 0.5 to 2 THzregion.
The detector elements are 3 mm diameter, accurately spaced on 5 mm centers.
Its high responsivity and very low noise level allow precise detection of weaksignals.

CUSTOM / OEM PRODUCTS

8 -CHANNEL PYROELECTRIC ARRAY
0.5 TO 2 THZ RANGEHIGH VOLTAGE RESPONSIVITYPRODUCT GUIDE 2024

T 418.651.8003 | info@gentec-eo.com

BEAM PROFILING

Gentec-EO is the only supplier able to manufacture beam dumps able to withstandthe tremendous peak power of a 10-petawatt laser, in a high vacuum environment.

.com

181

QUAD

Position sensing power & energy detectors

COMPATIBLE DISPLAYS & PC INTERFACESQUAD-4Track module

KEY FEATURESACCESSORIESMEASURE, TRACK AND ALIGN
With µm resolution in real time!
4-CHANNEL DETECTORS
Unique pyrolectric QUADrant detector
technology handles high peak power without
saturation

Stand with delrin post

Additional 9V power supply

SDC-500 digital
optical chopper (for -P)

Pelican carrying case	FOR CW, PULSED AND HIGH REP RATELASERS• QUAD-E: Energy per pulse from μJ to mJ• QUAD-P: Powers from μW to mW

FROM UV TO FIR AND THZBroadband detectors cover the full spectrum,from UV to Sub-Millimeter wavelengths
LARGE AREA SENSORS
9 mm and 20 mm square detectors
INCLUDES APPLICATION SOFTWARE
Complete LabVIEW application software
included, with many features

182

gentec-eo.com/customization

USB cable

QUADSpecifications

QUAD-9-MT-P

QUAD-20-MT-E

QUAD-20-MT-P

MAX ENERGY / AVG POWER

20 mJ

200 mW

20 mJ

200 mW

MAX POSITION RESOLUTION

1 µm

10 µm

1 µm

10 µm

EFFECTIVE APERTURE

9 x 9 mm

9 x 9 mm

20 x 20 mm

20 x 20 mm

Spectral range

0.1 - 3000 µm

0.1 - 3000 µm

0.1 - 3000 µm

0.1 - 3000 µm

Min beam size a≥ 4.5 mm Ø

≥ 4.5 mm Ø

≥ 10 mm Ø≥ 10 mm Ø

Position resolution with QUAD-4TRACK

1 µm

10 µm

1 µm

10 µm

Maximum measurable energy/power

20 mJ/channel

200 mW

20 mJ/channel

200 mW

Noise equivalent energy/power

0.5 µJ

1 µW

1.0 µJ

2 µW

Rise time (typical 0-100 %)150 µs< 0.02 s

150 µs< 0.02 s

Max repetition rate

1000 Hz

N/A

1000 Hz

N/A

Maximum pulse width

2.5 µs

N/A

2.5 µsN/A

Maximum chopping frequency

N/A

50 Hz

N/A

50 Hz

Sensitivity

1000 V/J

2000 V/W

1000 V/J

2000 V/W

POWER DETECTORSQUAD-9-MT-E

MEASUREMENT CAPABILITY
ENERGY DETECTORSBEAM PROFILINGDAMAGE THRESHOLDS
Max average power density (at 1064 nm)

100 mW/cm2

100 mW/cm2

100 mW/cm2

100 mW/cm2

Max energy density (at 1064 nm 10 ns)

50 mJ/cm

50 mJ/cm

2

50 mJ/cm

50 mJ/cm2

Effective aperture

9 x 9 mm

9 x 9 mm

20 x 20 mm

20 x 20 mm

Sensor type

Pyroelectric

Pyroelectric

Pyroelectric

Pyroelectric

Absorber

MT

MT

MT

MT

Dimensions

63.5Ø X 40.6D mm

63.5Ø X 40.6D mm

63.5Ø X 40.6D mm

63.5Ø X 40.6D mm

Weight

181 g

181 g

181 g

181 g

STAND-D-233

STAND-D-233

STAND-D-233

STAND-D-233

2

2

PHYSICAL CHARACTERISTICSTERAHERTZ DETECTORSORDERING INFORMATION
Compatible standProduct pageDISPLAYS & PC INTERFACES

a. For optimal performance.

CUSTOM / OEM PRODUCTS

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

183

QUAD
Specifications

QUAD-4TRACKThe QUAD-4Track is a laser position sensing system designed to support ourunique pyroelectric quadrant detectors, QUAD-P and QUAD-E. It is a 4-channel
microprocessor-based system that measures the voltage output of each QUAD
element and does the math necessary to provide a measurement of the X and Y
displacement of a laser beam or image. It is fast and can be used to track, alignand/or measure movement in real time, with a resolution of just a few microns!

SPECIFICATIONS & FEATURES
QUAD-4Track(front view)

QUAD-4TRACK
Number of channels

4Full scale ranges (4 decades) (E / P)

QUAD-4Track(rear view)

Energy mode (with QUAD-E)

20 µJ to 20 mJ

Power mode (with QUAD-P)

200 µW to 200 mW

USB connection to computer

YES (USB 2.0 full speed)

Power supply

9 VDC

Power on light

YES

Detector input

DB-25 connector

Detector analog output

BNC connector (0 - 2 V)

Trigger input (TTL)

BNC connector with LED indicator

Product number

201517QUAD DETECTORS
Our large area pyroelectric quadrant detectors provide unique advantages overother position sensing detectors like silicon quads or lateral effect photodiodes.
They are fast, handle high peak power of pulsed lasers without saturation andrespond to lasers across the spectrum, from UV to Far IR and even THz. The
QUAD-E is intended for use with pulsed sources at up to 1000 Hz, while the
QUAD-P is designed for CW and High Repetition Rate (Quasi-CW) sources. Bothtypes of detectors can also be used as standalone units, in an analog mode, forincorporation into your own system application. We can provide a Lemo cable forthis purpose.

ANALOG OUTPUT
The analog output of the QUAD-4Track provides voltage that is directly proportional
to the pulse energy or laser power irradiating each QUAD element. When thefour voltage outputs are equal, the beam is centered on the QUAD detector. This
provides a very useful tool when setting up our QUAD probes with your source for
optical alignment.

184

gentec-eo.com/customization

QUADSpecifications

MEASUREMENT SCREENDATA LOGGINGPRODUCT GUIDE 2024

T 418.651.8003 | info@gentec-eo.com.com

CUSTOM / OEM PRODUCTS

Another very handy feature is “data logging”. This allows you to set up the QUAD4Track to follow the displacement, energy and/or power of your laser over several
minutes, hours or even days. Need to measure the “beam steering” of your laser as itwarms up? This is how you do it! Need to measure the beam displacement vs laser
repetition rate or energy level? Data logging will help you measure it!

DISPLAYS & PC INTERFACES

We’ve developed a unique position calibration routine which allows you to calibrate
our QUAD-4Track system when working with a uniformly round laser beam. It
requires the use of a micrometer-driven linear stage (1-axis only). As you can see fromthe calibration screen on the left, the procedure involves zeroing the instrument,
moving the QUAD probe to nine discrete positions (+2.000 to -2.000 mm) andthen capturing the QUAD readings. It then determines correction coefficients (lastcolumn) and applies them to the raw data to arrive at “corrected positions”. The
QUAD probe is now calibrated!

TERAHERTZ DETECTORSPOSITION CALIBRATION SCREEN

BEAM PROFILING

In the measurement screen shown on the left, we are tracking the beam stabilityof a pulsed Nd:YLF laser at 10 Hz. The resolution was set at 0.001 µm, the boundaryis at 20 µm (red circle), and the zoom feature is at 64X. The total energy is 108.5 µJ,
the final position of the laser is at -8 µm in X and -8 µm in Y. The green tracking line
shows the movement of the laser about the zero position over a few hundred pulses.

ENERGY DETECTORSTRACKING THE BEAM OVER TIME

POWER DETECTORSQUAD-4Track includes powerful, stand alone, LabVIEW software which is used
to control the instrument, process the data, and display X and Y position. It also
displays the energy or power of your source and repetition rate. The large graphic inthis screen shows the position of the centroid of the beam and tracks its movement
in real time. The software includes many handy features like: set boundary, zoom (2X
to 128X), set resolution, data logging, and many more. The green line represents the
tracking history.

185

STEP

5 nW - 0.5 mW, temperature-controlled pyroelectric detectors
for ultra-stable laser power measurements
COMPATIBLE DISPLAYS & PC INTERFACESPlug your TP sensor into the STEP-Controller for power supply andtemperature control. You can then use the analog output with a scope or
lock-in

STEP-Controller

KEY FEATURESACCESSORIESSPECTRALLY FLAT RESPONSE
These radiometers were developed for NIST, to be
used with a broadband spectrometer to act as aspectral transfer standard when calibrating otherdetectors in the 0.25 to 15 μm range.
TEMPERATURE-CONTROLLED
POWER MEASUREMENT
Each head is composed of a low noise detector,
thermistor, TE cooler and heatsink to compensate
for any temperature change
THE ULTIMATE CHOICE IN
MEASUREMENT STABILITY
Temperature control down to 0.05°C from 20 to
30°C gives a temperature coefficient < 0.01 %, thus
a voltage output stable to 0.01 %2 SIZES AVAILABLE• TP5-BL: 5 mm Ø pyroelectric sensor with organic
black coating• TP9-BL: 9 mm Ø pyroelectric sensor with organic
black coating

186

gentec-eo.com/customization

Stand with delrin post

Removable IR windows(Various types available)

SDC-500 digital
optical chopper (for -P)

Pelican carrying case

Fiber adaptors & connectors(FC, SC, ST and SMA)STEPSpecifications

TP9-BLMAX AVERAGE POWER

0.5 mW

0.5 mW

EFFECTIVE APERTURE

5 mm Ø

9 mm Ø

TEMPERATURE STABILITY± 0.05°C± 0.05°C

Spectral rangea

0.25 - 15 µm

0.25 - 15 µm

Temperature stability± 0.05°C± 0.05°C

Voltage response stability± 0.01 %± 0.01 %Maximum average power

0.5 mW

0.5 mW

Noise equivalent powerb< 5 nW< 5 nW

Rise time≤ 0.2 s≤ 0.2 sCalibration uncertainty±2.5 % (633 nm)±2.5 % (633 nm)

Chopper frequencyc

10 Hz, 50 % duty cycle

10 Hz, 50 % duty cycle

50 mW/cm2

50 mW/cm2

Effective aperture

5 mm Ø

9 mm Ø

SensorPyroelectric

Pyroelectric

Absorber

BL

BL

Dimensions

50.8Ø x 48.3D mm

50.8Ø x 48.3D mm

Weight0.23 kg

0.23 kg

Available output options

LEMO only

LEMO only

Compatible stand

STAND-D-233

STAND-D-233

POWER DETECTORS

TP5-BL

MEASUREMENT CAPABILITY
ENERGY DETECTORSDAMAGE THRESHOLDSBEAM PROFILING

Max average power density (at 1064 nm)
PHYSICAL CHARACTERISTICSTERAHERTZ DETECTORSORDERING INFORMATION

Product pageDISPLAYS & PC INTERFACES

a. NIST-traceable calibration at 632.8 nm.
b. With STEP-Controller and scope. Noise is < 11 nW with STEP-controller and Lock-In.
c. SDC-500 optical chopper sold separately.

CUSTOM / OEM PRODUCTS

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

187

TRAP

Optical trap detectors

COMPATIBLE DISPLAYS & PC INTERFACES
TRAP -PREAMP
Low noise, high gain amplifier

KEY FEATURESACCESSORIESHAVE YOUR OWN GOLDENCALIBRATION STANDARD
The high quantum efficiency (>99 %) makes it
an excellent calibration transfer standard
INCREDIBLE SPATIAL UNIFORMITY
The spatial uniformity is better than 0.05 %LOW CALIBRATION UNCERTAINTY
From 440 to 980 nm
FOR DIVERGENT OR COLLIMATED BEAMS
Devices optimized for both types of lasers
MEASURE POWER FROM NW TO MW
When used with the TRAP-PREAMP amplifierthat provides a direct digital readout

188

gentec-eo.com/customization

Stand with delrin post

Pelican carrying case

TRAPSpecifications

TRAP7-SI-D-BNC

MAX AVERAGE POWER

1 mW

1 mW

EFFECTIVE APERTURE

7 mm Ø

7 mm Ø

BEAM TYPE

CollimatedDivergent

Spectral range

200 - 1100 nm

200 - 1100 nm

Calibrated spectral range

440 - 980 nm

440 - 980 nm

Maximum average power

1 mW

1 mW

Noise equivalent power

100 pW

100 pW

Current responsivity (at 630 nm)

0.5054 A/W

0.5054 A/W

Beam type

CollimatedCollimated or divergent

Field of view

± 10°

± 14°

Quantum efficiency> 99 %> 99 %

Calibration uncertainty< ±1.0 % (440 - 980 nm)< ±1.0 % (440 - 980 nm)

Spatial uniformity

0.05 % in 5 mm diameter0.05 % in 5 mm diameter

1 mW/cm2

1 mW/cm2

Aperture diameter

7 mm Ø

7 mm Ø

SensorSiliconSilicon

Dimensions

69Ø x 27.7D mm

69Ø x 27.7D mm

Weight0.23 kg

0.23 kg

Available output options

BNC only

BNC only

Compatible stand

STAND-D-233

STAND-D-233

POWER DETECTORS

TRAP7-SI-C-BNC

MEASUREMENT CAPABILITY
ENERGY DETECTORSBEAM PROFILINGDAMAGE THRESHOLDS
Maximum average power density
PHYSICAL CHARACTERISTICSTERAHERTZ DETECTORSORDERING INFORMATION

Product pageDISPLAYS & PC INTERFACESCUSTOM / OEM PRODUCTS

PRODUCT GUIDE 2024

Specifications are subject to change without notice
T 418.651.8003 | info@gentec-eo.com

.com

189

DISTRIBUTORS WORLDWIDECountries with a calibration center

Countries with a Gentec-EO office

AMERICAS
UNITED STATES & CANADAGENTEC ELECTRO-OPTICS, INC.

Félicien Legrandflegrand@gentec-eo.com

gentec-eo.com

SOUTH AMERICA

PHOTONICS LTDA

Ettore Cortese

info@photonics.com.br

photonics.com.br

FRANCELASER COMPONENTS S.A.S.

Elvyne Egrot

e.egrot@lasercomponents.fr

lasercomponents.com

GERMANY, AUSTRIA, CZECHIA & POLAND

LASER COMPONENTS GMBH

Nadine Kujathn.kujath@lasercomponents.com

lasercomponents.com

GREECEOPTRONICS TECHNOLOGIES S.A.

Nina Christodoulia

nchristodoulia@optronics.gr

optronics.gr

ITALY

OPTOPRIM S.R.L.

Daniele Turiano

turiano@optoprim.it

optoprim.it

LITHUANIA, ESTONIA, BELARUS & LATVIASTANDA

Jonas Kniuksta

sales@standa.lt

standa.lt

NETHERLANDS, BELGIUM & LUXEMBOURG TE LINTELO SYSTEMS BV

Roland Kuijvenhoven

roland@tlsbv.nl

tlsbv.nl

SPAIN & PORTUGALLuz Ruizluz.ruiz@pro-lite.es

pro-lite.es

SWEDEN, DENMARK, FINLAND & NORWAY LASER COMPONENTS NORDIC AB

Harvey Washbrookh.washbrook@lasercomponents.se lasercomponents.com

SWITZERLAND

GMP SA

Pablo Moreno

pablo.moreno@gmp.ch

gmp.chTURKEYTLS LAZER

Mustafa Ersin

m.ersin@tlslaser.com

tlslaser.com

UNITED KINGDOM & IRELAND

LASER LINES LTD

Dan Cort

danc@laserlines.co.uk

laserlines.co.uk

AUSTRALIA & NEW ZEALANDLASTEK PTY LTD.

Jessica Mackintosh

sales@lastek.com.au

lastek.com.au

CHINA & HONG KONG

OPCROWN PHOTONICS CO., LTD.

Gloria Yanggloria.yang@opcrown.com

opcrown.com

INDIA

SOLUTIONS FOR LASERS & PHOTONICS
PVT LTD.

Abhishek Losalka

info@lasersandphotonics.com

lasersandphotonics.com

JAPAN

GENTEC-EO JAPAN GK

Daisuke Satodsato@gentec-eo.com

gentec-eo.com

KOREA

QBIC LASER SYSTEM, INC.

Changkon Kim

sales@qbiclaser.com

qbiclaser.com

MALAYSIADESEO ENTERPRISE

Eddy Looi

sales@deseoent.com

deseoent.com

RUSSIAN FEDERATIONETM PHOTONICS LLC

Vladimir Shershulin

sales@etm-p.ru

etm-p.ru

SINGAPORE & PHILIPPINESACEXON TECHNOLOGIES PTE LTD

Lawrence Chuasales@acexon.com

acexontech.com

TAIWANONSET ELECTRO-OPTICS

Patrick Huangsales@onset-eo.com

onset-eo.com

THAILANDWETLING CO., LTD

Phakkawat Attaphiphatwanich sales@laser-gen.com

laser-gen.com

VIETNAMATEK PHOTONICS CO., LTD

Nguyen Tat Thanh

thanh.nt@atekvietnam.com

atekvietnam.com

ISRAELLAHAT TECHNOLOGIES LTD.

Yamit Avni

yamit.avni@lahat.com

lahat.com

UNITED ARAB EMIRATESARABIAN CALIBRATION DEV. REP. CO. LLC Jude Dias

sourcing@arabcal.com

arabcal.com

PAKISTANCOMPREHENSIVE SOLUTIONS

Zeeshan Akram

zeeshan@comprehensive-sol.com

comprehensive-sol.com

GENTEC ELECTRO-OPTICS, INC.

Charles Dumas

cdumas@gentec-eo.com

gentec-eo.com

EUROPE

PRO-LITE TECHNOLOGY IBERIA, S.L.

ASIA-PACIFIC

MIDDLE EAST - AFRICA

ALL OTHER COUNTRIESALL OTHER COUNTRIES

190

gentec-eo.com

RETURNS & WARRANTYCALIBRATION AND REPAIR SERVICEAll Gentec-EO products receive a NIST-traceable calibration and are shipped with a Calibration Certificate to prove it. The certificate
tells you the sensitivity of your power or energy head, the ambient calibration conditions, and a list of all the NIST-traceable standards
and instruments used in the calibration.
The actual need for recalibration depends on use and environmental conditions. Under typical operating conditions and laser
exposures annual recalibration is the industry standard recommended by calibration experts such as NIST. Our highly professional
service department is happy to recalibrate or repair your instrument any time you need it. In every case, you will get the same accurate
calibration and detailed certificate as when your instrument was new. In addition, we do an incoming calibration test to let you know
how the device was performing before service. We will help you meet any ISO and quality requirements. Here is how to send an RMA
request:BY PHONEBY E-MAIL

Mr. Nicolas Litalien
1-418-651-8003 ext. 302

Mr. Nicolas Litalien
service@gentec-eo.com

USING OUR ONLINE RMA FORM
Go to https://www.gentec-eo.com/contact-us/support-rma-request
Fill out the online form and click “SUBMIT MY REQUEST”

IN ALL CASES, PLEASE PREPARE THE FOLLOWING INFORMATION BEFORE CONTACTING US:• Model name(s)• Serial number(s)• If a repair is needed, please provide a description of the problem

PRODUCT GUIDE 2024

T 418.651.8003 | info@gentec-eo.com

.com

191

DOWNLOAD THE
DIGITAL VERSION:
WWW.GENTEC-EO.COM

LEADER IN LASER BEAM
MEASUREMENT SINCE 1972
CANADA

UNITED STATESJAPANCALIBRATION CENTERS

445 St-Jean-Baptiste, Suite 160
Quebec, QC, G2E 5N7, Canada

5825 Jean Road Center
Lake Oswego, OR, 97035, USA

445 St-Jean-Baptiste, Suite 160
Quebec, QC, G2E 5N7, Canada

T(418) 651-8003

T(503) 697-1870

Office No. 101, EXL111 building,
1-1-1, Takinogawa, Kita-ku, Tokyo
114-0023, JapanF(418) 651-1174

F(503) 697-0633

info@gentec-eo.com

info@gentec-eo.com

T+81-3-5972-1290

F+81-3-5972-1291
info@gentec-eo.com

Werner von Siemens Str. 15
82140 Olching, Germany
Office No. 101, EXL111 building,
1-1-1, Takinogawa, Kita-ku, Tokyo
114-0023, Japan