FC400-CCL-FA
FC400-DAC-FA
FC400-EIP-FA
FC400-ECT-FA
FC400-232-FA
DIN-RAIL MOUNT DIGITAL INDICATOR
CATOR
Applicable from fast response measurement
to multi-weighing with high accuracy!
FC400-CCL-FA
FC400-DAC-FA
FC400-EIP-FA
FC400-ECT-FA
FC400-232-FA

A measurement system can be configured easily using CC-Link network.
Either voltage or current output, corresponding to the indicated value, is available.
A measurement system can be configured easily using EtherNet/IP network.
A measurement system can be configured easily using EtherCAT network.
Measured data and results can be collected using RS-232C network.

High sampling rate & resolution

Various interfaces

High-Speed A/D conversion and powerful digital processing
capability of 2400 times/sec.
High display resolution of 1/100000 (max).

FC400-CCL-FA
FC400-DAC-FA
FC400-EIP-FA
FC400-ECT-FA
FC400-232-FA

Equipped with CC-Link, SI/F, USB interface
Equipped with D/A converter, RS-485, USB interface
Equipped with EtherNet/IP and USB interface
Equipped with EtherCAT and USB interface
Equipped with RS-232C, SI/F, USB interfaceex) CC-Link
EtherNet/IP
EtherCATRS-485

Measurement can be performed quickly andprecisely due to high speed A/D conversion.

With communication through USB
interface, logging, graph display,setting parameters, and calibration
can be done.

High performance filter & Auto filter adjustment

By simply
plypressingg
the autooadjustment
ent
key….

CC-Link
EtherNet/IP
EtherCAT
RS-485

Incomplete

Complete

With combination of low pass filter (0.1 to 600 Hz) whichcorresponds to various vibration and moving average filter(OFF, 2 to 999) that is effective for periodic vibration,
automatic searching for optimal value with accuracy andstability can be achieved.
(You can manually adjust settings too)The
readings
fluctuate….

FC400

Sensorinput

Application software for USB interface

Press Press Caulking Pressex) CC-Link
EtherNet/IP
EtherCAT
RS-485
RS-232C
D/A converter
SI/F

PLC

Sensor
input
SI/F

Display etc.

D/A
FRQYHUWHU Recorder

Stabilized!
Stabiliz

RS-232C

Compactt sizee

CC-Link
EtherNet/IP
EtherCAT
RS-485FC400

It can be fit into a limited space!
Light weight & compact body for 35 mm DIN rail mount.
ex) USB

PLC

FC400
USBPC

Digital peak & bottom hold, Hi & Lo function

Input conversion value display

Quickly detect high speed changes & make OK / NOK judgement

The output signal level of load cell can be displayed in mV/V/ formonitor purpose.
Malfunction indicator or faulty sensor can be differentiated easily.

Equivalent input calibration

6-digit display

Theoretical calibration can be performed easily by registering thecapacity and rated output of load cells.

24 bit A/D converter enables 6-digit display

Specifications
AnalogDisplay

HoldExternalsignalExcitation voltage

DC 5 V±5% Output current: Within 90 mADC 2.5 V±5% Output current: Within 45 mA(depending on settings)
Ratiometric method(Up to 6 350 Ω load cells can be connected in parallel)
Signal input range-3.9 to +3.9 mV/V
Zero adjustment range Automatic adjustment by digital processing -3.0 to +3.0 mV/V
Span adjustment range Automatic adjustment by digital processing 0.01 to 3.8 mV/V
Min. input sensitivity
0.15 μV/countAccuracyNon-linearity: Within 0.01% FS
Zero drift:
0.0002% FS/Υ typ. * When calibrated by 3 mV/V with 5 Vof the excitation voltage
Gain drift:
1 ppm/Υ typ.
FilterDigital low-pass filter: 0.1 to 600 Hz
Moving average filter: OFF, 2 to 999 times
A/D converter
Speed:
2400 times/sec.
Resolution: 24 bit (binary)
Display unit
Character height: 8 mm
Numerical display by 7-segment green LED
Display value
Up to 6 digits
Display frequency
Selectable from 1, 3, 6, 13, 25 times/sec.
Status display
RUN, SD, RD, ERR (FC400-CCL-FA)
MS, NS (FC400-EIP-FA)
RUN, ERR (FC400-ECT-FA)
Sample, Peak, Bottom, Peak&Bottom, Average
Output signal
Transistor’s open collector output(5)
Vceo = 30 V, Ic = 50 mA
Input signal
Selectable/configurable(3)
Input is ON when shorted to COM terminal by contact (relayswitch, etc.) or non contact (transistor, open collector, etc.).
* DC 24 V external power supply is required.

Interface CC-Link interface (FC400-CCL-FA)
D/A converter interface (Voltage and current output) (FC400-DAC-FA)
EtherNet/IP interface (FC400-EIP-FA)
EtherCAT interface (FC400-ECT-FA)
RS-232C interface (Selectable from Modbus-RTU or UNI-Format) (FC400-232-FA)
RS-485 interface (Selectable from Modbus-RTU or UNI-Format) (FC400-DAC-FA)
SI/F 2-wire type serial interface (FC400-CCL-FA, FC400-232-FA)
USB interface

Power supply voltage DC 24 V (±15%)
General
3 W typ. (FC400-DAC-FA, FC400-232-FA)
specification Power consumption
4 W typ. (FC400-CCL-FA, FC400-EIP-FA, FC400-ECT-FA)
Operating conditions Operating temperature range: -10 to +50Υ
Storage temperature range: -20 to +85Υ
Humidity: 85% RH or less (non-condensing)34(W) × 88(H) × 91(D) mm (Projections excluded)
Dimensions
Approx. 210 g (FC400-CCL-FA, FC400-DAC-FA)
WeightApprox. 230 g (FC400-EIP-FA, FC400-ECT-FA)
Approx. 220 g (FC400-232-FA)
Attachment Quick manual…2
Jumper wire…2
Mini driver…1
CC-Link connector (FC400-CCL-FA)…1
Various I/O connector (FC400-CCL-FA, FC400-EIP-FA, FC400-ECT-FA)…2(FC400-DAC-FA, FC400-232-FA)…3
Accessories CA81-USB: miniUSB-computer USB cable 1.8 m
CN74:
CC-Link connector (Same accessory as the attached one)
CN75:
CC-Link connector (Y type branch connector)
CN76:
CC-Link connector (Terminator connector)
CN85:
13 p connector for power source/sensor/RS-485(Same accessory as the attached one)CN86:
3 p connector D/A converter (Common with RS-232C connector)(Same accessory as the attached one)CN87:
10 p connector for external I/O (Same accessory as the attached one)
CE marking EMC directive EN61326-1
certification* FC400-ECT-FA is Free Run mode.
* Please note that there are possibilities of individual differences in a color tone on display devices
such as LEDs, fluorescent display tubes and LCDs due to manufacturing process or production lots.

Structure of product code
Model
FC400-CCL-FA
FC400-DAC-FA
FC400-EIP-FA
FC400-ECT-FA
FC400-232-FA

Interface
CC-Link, SI/F, USB
D/A converter (Voltage and current output), RS-485, USB
EtherNet/IP, USB
EtherCAT, USB
RS-232C, SI/F, USB

External dimension

91

34

DIN-rail mountadapter

88
101

DAC-FA

91

34

DIN-rail mountadapter

88
101

EIP-FA‫ ڦ‬FC400-ECT-FA

88
101

EIP

‫ ڦ‬FC400-EIP-FA

34‫ ڦ‬FC400-232-FA

88
101

DAC

CCL-FA‫ ڦ‬FC400-DAC-FA

88
101

CCL

‫ ڦ‬FC400-CCL-FA

91

DIN-rail mountadapter

34

91

DIN-rail mountadapter

34

91

DIN-rail mountadapter