PSA Oxygen Generator

Centrox
The Centrox is an oxygen generator ideal foruse in many applications. The Centrox’s ability
to deliver up to 95% oxygen concentration
at flows of up to 32 SCFH (0.84 Nm3/hr or15 LPM) make it a great fit for medical uses
requiring up to 50 psig (345 kPa or 3.5 barg).

Typical Applications■ Medical Oxygen Supply■ Veterinary Medicine

Note: The 220 V ~ ±10%, 50 Hz configuration is		for medical use outside of the USA.

Features
PSA Generator Module■ Self-contained (internal air compressors)■ Microprocessor controlled■ Low operating cost
■ Automatic and unattended operation■ Wall or rack mount design■ Heavy duty metal enclosures■ Oxygen concentration monitor

Air Compressor Module

Specifications
Product Flow1

32 SCFH (0.84 Nm3/hr or 15 LPM)

Product Pressure

50 psig (345 kPa or 3.5 barg)

Product Concentration

Up to 95%

Product Dew Point-100°F (-73°C)

Dimensions (W x D x H) (Nominal)■ Air Compressor Module■ PSA Generator Module

21 x 13 x 27 in (52 x 32 x 68 cm)
27 x 15 x 38 in (68 x 37 x 97 cm)

Weight■ Air Compressor Module■ PSA Generator Module

100 lb (45 kg)
160 lb (73 kg)

Physical Connection
Product Gas Outlet

B size oxygen demand valve

Ambient Operating Conditions

Locate the oxygen generator in a well-ventilated area that is protected from weather elements and remains between 40°F
(4°C) and 112°F (44°C)

Control Power Requirements (Single Phase)

220 V ~ ±10%, 50 Hz, 5.5 A

60 Gallon Oxygen Receiver

1Dimensions (Dia. x H)

26 x 52 in (65 x 131 cm)

Weight159 lb (72 kg)

SCF (Standard cubic foot) gas measured at 1 atmosphere and 70°F / Nm3 (Normal cubic meter) gas measured at 1 atmosphere and 0°C / LPM (Liters per minute) gas measured at 1 atmosphere and 21ºC

60 Gallon Receiver

Note: All dimensions are nominal.

ModelPart NumberCentrox1

AS074-2

220 V ~ ±10%, 50 Hz with oxygen concentration monitor (LED indicator)

TA150-1

60 Gallon Oxygen Receiver

KI046-1

Accessory Kit (10 ft Stainless Braided Oxygen Hose & Primary/Secondary Oxygen Ball Valve Assembly)

KI470-1

Accessory Kit (10 ft Stainless Braided Oxygen Hose & Regulator Assembly with Flowmeter)

KI225-1

Accessory Starter Kit(FOR EXPORT ONLY)

RequiredAccessories

OptionalAccessoriesDescription

Ordering Information

TA212-1

Accessory Kit, Mini Oxygen Receiver

AN021-1

Oxygen Analyzer (Maxtec Handi)

AN005-1

Oxygen Analyzer (Maxtec Max O2 Plus)

AN075-1

Oxygen Analyzer/Sensor (Maxtec Max O2 Plus)

Shipping Information

Centrox

60 Gallon Oxygen Receiver

Class

92.5

70

Commodity Classification Number

8421.39.8040

7311.00.0000

Dimensions (W x D x H)

34 x 52 x 50 in (86 x 132 x 127 cm) with Accessory Kits, Crated

28 x 30 x 56 in (71 x 76 x 142 cm)

Gross Weight559 lb (254 kg) with Accessory Kits, Crated

177 lb (80 kg)

Warranty: 1 Year Parts and Factory Labor** An unprotected or inadequately ventilated environment, or improper control power may cause damage to the oxygen generator
not covered under warranty.
1Interconnecting red compressed air hose included.
All performance ratings based on an ambient temperature up to 100°F (38°C), up to 1,000 feet elevation, and 80% relative humidity.

MADE IN USA© 2020 AirSep Corporation. All Rights Reserved.
AirSep reserves the right to discontinue its products, or change the prices, materials, equipment, quality, descriptions,specifications and/or processes to its products at any time without prior notice and with no further obligation orconsequence. Any rights expressly stated herein are reserved by us, as applicable.

260 Creekside Drive
Buffalo, NY 14228-2075 U.S.A.
Tel: (716) 691-0202 ■ Fax: (716) 691-1255
www.airsep.com ■ info@airsep.com
ML-IND0058 A