PSA Oxygen Generator

Reliant

The Reliant is an oxygen generator ideal foruse in many applications. The Reliant’s ability todeliver up to 95% oxygen concentration at flowsof up to 17 SCFH (0.44 Nm3/hr or 8 LPM) makeit a great fit for medical clinics requiring up to 50
psig (345 kPa or 3.45 barg).

Typical Applications■ Medical Oxygen Supply■ Veterinary Medicine

Note: The 220 V ~ ±10%, 50 Hz configuration is		for medical use outside of the USA.

Features■ Self-contained (internal air compressor)■ Microprocessor controlled■ Low operating cost■ Automatic and unattended operation■ Transportable design■ Easy to install and maintain■ Oxygen concentration monitor and alarm■ Optional auxiliary oxygen receiver(free standing)Specifications
Product Flow

1

17 SCFH (0.44 Nm /hr or 8 LPM)
3

Product Pressure

50 psig (345 kPa or 3.45 barg)

Product Concentration

Up to 95%

Product Dew Point-100°F (-73°C)

Dimensions (W x D x H) (Nominal)

22.5 x 24.5 x 36.25 in (57.2 x 62.3 x 92.1 cm)

Weight175 lb (79 kg)

Physical Connections
Product Gas Outlet

B size oxygen demand valveSound Level

58 dBA @ 1 meter, open field conditions

Ambient Operating Conditions

Locate the oxygen concentrator in a well-ventilated area that is protected from weather elements and remains between 40°F
(4°C) and 112°F (44°C)

Control Power Requirements (Single Phase)

220 V ±10%, 50 Hz, 3.0 A

Typical Power Consumption

680 W

60 Gallon Oxygen Receiver (Optional)

1Dimensions (Dia. x H)

26 x 52 in (65 x 131 cm)

Weight159 lb (72 kg)

SCF (Standard cubic foot) gas measured at 1 atmosphere and 70°F / Nm3 (Normal cubic meter) gas measured at 1 atmosphere and 0°C / LPM (Liters per minute) gas measured at 1 atmosphere and 21ºC

22.5
[57.2]

36.25[92.1]

24.5[62.3]

Note: All dimensions are nominal.

Optional 60 Gallon Receiver

Ordering Information
ModelPart Number

ReliantDescription

AS072-3

220 V ~ ±10%, 50 Hz with oxygen concentration monitor (LED indicator)(FOR EXPORT ONLY)

AS072-5

220 V ~ ±10%, 50 Hz with oxygen concentration monitor (LED indicator), Russian Language

RequiredAccessory

KI046-1

Accessory Kit (10 ft Stainless Braided Oxygen Hose & Primary/Secondary Oxygen Ball Valve Assembly)

KI048-1

Accessory Kit (10 ft Stainless Braided Oxygen Hose & Oxygen Receiver Purge Ball Valve)

Optional
Accessories

KI168-1

Accessory Starter Kit

TA150-1

60 Gallon Oxygen Receiver

AN021-1

Oxygen Analyzer (Maxtec Handi)

AN005-1

Oxygen Analyzer (Maxtec Max O2 Plus)

AN075-1

Oxygen Analyzer/Sensor (Maxtec Max O2 Plus)

Shipping Information

Reliant

60 Gallon Oxygen ReceiverClass92.5

70

Commodity Classification Number

8421.39.8040

7311.00.0000

Dimensions (W x D x H)

28 x 30 x 45 in (71 x 76 x 114 cm) with Accessory Kit

28 x 30 x 56 in (71 x 76 x 142 cm)

Gross Weight

233 lb (106 kg) with Accessory Kit

177 lb (80 kg)

Warranty: 1 Year Parts and Factory Labor** An unprotected or inadequately ventilated environment, or improper control power may cause damage to the oxygen generator
not covered under warranty.
All performance ratings based on an ambient temperature up to 100°F (38°C), up to 1,000 feet elevation, and 80% relative humidity.

MADE IN USA© 2020 AirSep Corporation. All Rights Reserved.
AirSep reserves the right to discontinue its products, or change the prices, materials, equipment, quality, descriptions,specifications and/or processes to its products at any time without prior notice and with no further obligation orconsequence. Any rights expressly stated herein are reserved by us, as applicable.

260 Creekside Drive
Buffalo, NY 14228-2075 U.S.A.
Tel: (716) 691-0202 ■ Fax: (716) 691-1255
www.airsep.com ■ info@airsep.com
ML-IND0057 A