GP01211D/06/EN/01.23-00
71637048
2024-03-15

Products

Solutions

Services

Valid as of version
01.00.zz (Device firmware)

Description of Device Parameters
Proline Teqwave M 300
Total solids measurement via microwave transmission
HART

Proline Teqwave M 300 HART

Table of contents

Table of contents
1

About this document . . . . . . . . . . . . . . . . 4

1.1
1.2
1.3

Document function . . . . . . . . . . . . . . . . . . . . . 4
Target group . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Using this document . . . . . . . . . . . . . . . . . . . . 4
1.3.1 Information on the document
structure . . . . . . . . . . . . . . . . . . . . . . . 4
1.3.2 Structure of a parameter description . . 6
Symbols used . . . . . . . . . . . . . . . . . . . . . . . . . . 6
1.4.1 Symbols forcertain types of information . . . . . . . . . 6
1.4.2 Symbols in graphics . . . . . . . . . . . . . . . 7
Documentation . . . . . . . . . . . . . . . . . . . . . . . . 7
1.5.1 Standard documentation . . . . . . . . . . . 7
1.5.2 Supplementary device-dependent
documentation . . . . . . . . . . . . . . . . . . 7

1.4

1.5

3.8.4

"Main electronic module + I/O
module 1" submenu . . . . . . . . . . . . . 161
3.8.5 "Sensor electronic module (ISEM)"
submenu . . . . . . . . . . . . . . . . . . . . . 162
3.8.6 "I/O module 2" submenu . . . . . . . . . . 163
3.8.7 "I/O module 3" submenu . . . . . . . . . . 164
3.8.8 "Display module" submenu . . . . . . . . 165
3.8.9 "Data logging" submenu . . . . . . . . . . 166
3.8.10 "Min/max values" submenu . . . . . . . 174
3.8.11 "Heartbeat Technology" submenu . . . 178
3.8.12 "Simulation" submenu . . . . . . . . . . . 179

4Country-specific factory settings . . 188

4.1

SI units . . . . . . . . . . . . . . . . . . . . . . . . . . . .
4.1.1 System units . . . . . . . . . . . . . . . . . .
4.1.2 Output current span . . . . . . . . . . . .
US units . . . . . . . . . . . . . . . . . . . . . . . . . . .
4.2.1 System units . . . . . . . . . . . . . . . . . .
4.2.2 Output current span . . . . . . . . . . . .

188
188
188
188
188
189

2

Overview of the Expert operating
menu . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8

4.2

3

Description of device parameters . . . 11

Index . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 190

3.1"System" submenu . . . . . . . . . . . . . . . . . . . . . 13
3.1.1 "Display" submenu . . . . . . . . . . . . . . . 13
3.1.2 "Configuration backup" submenu . . . . 32
3.1.3 "Diagnostic handling" submenu . . . . . . 35
3.1.4 "Administration" submenu . . . . . . . . . 41
"Sensor" submenu . . . . . . . . . . . . . . . . . . . . . . 45
3.2.1 "Measured values" submenu . . . . . . . . 46
3.2.2 "System units" submenu . . . . . . . . . . . 54
3.2.3 "Process parameters" submenu . . . . . . 57
3.2.4 "External compensation" submenu . . . 61
3.2.5 "Sensor adjustment" submenu . . . . . . . 62
3.2.6 "Factory adjustment" submenu . . . . . . 65
"I/O configuration" submenu . . . . . . . . . . . . . . 66
"Input" submenu . . . . . . . . . . . . . . . . . . . . . . . 68
3.4.1 "Current input 1 to n" submenu . . . . . . 68
3.4.2 "Status input 1 to n" submenu . . . . . . . 71
"Output" submenu . . . . . . . . . . . . . . . . . . . . . . 73
3.5.1 "Current output 1 to n" submenu . . . . . 74
3.5.2 "Pulse/frequency/switch output
1 to n" submenu . . . . . . . . . . . . . . . . 85
3.5.3 "Relay output 1 to n" submenu . . . . . 103"Communication" submenu . . . . . . . . . . . . . . 109
3.6.1 "HART input" submenu . . . . . . . . . . . 110
3.6.2 "HART output" submenu . . . . . . . . . . 115
3.6.3 "Diagnostic configuration" submenu . 131
3.6.4 "Web server" submenu . . . . . . . . . . . 135
3.6.5 "WLAN settings" wizard . . . . . . . . . . 139"Application" submenu . . . . . . . . . . . . . . . . . 146
3.7.1 "Totalizer 1 to n" submenu . . . . . . . . 146"Diagnostics" submenu . . . . . . . . . . . . . . . . . 150
3.8.1 "Diagnostic list" submenu . . . . . . . . . 153
3.8.2 "Event logbook" submenu . . . . . . . . . 155
3.8.3 "Device information" submenu . . . . . 157

3.2

3.3
3.4
3.5

3.6

3.7
3.8

Endress+Hauser

3About this document

Proline Teqwave M 300 HART

1

About this document

1.1

Document function

The document is part of the Operating Instructions and serves as a reference forparameters, providing a detailed explanation of each individual parameter of the operating
menu.

1.2

Target group

The document is aimed at specialists who work with the device over the entire life cycle
and perform specific configurations.

1.3

Using this document

1.3.1

Information on the document structure

The document lists the submenus and their parameters according to the structure from theExpert menu (→  8), which is displayed when the "Maintenance" user role isenabled.

4

Endress+Hauser

Proline Teqwave M 300 HART

About this document

Operating menu for operators and maintenances
Language
Operatation!
Language

Operator

Parameter 1
Parameter n
Submenu 1
Submenu n
Setup

Device tag

Wizard n / Parameter nAdvanced setup

Enter access code
Parameter 1

Task-oriented

Maintenance

Wizard 1 / Parameter 1

Parameter n
Submenu 1
Submenu n
Diagnostics

Parameter 1
Parameter n
Submenu 1
Submenu nOperating menu for experts
ExpertAccess status display
Parameter n

Sensor
Input
OutputCommunication

Function-oriented

ExpertSystemApplication
Diagnostics

A0029160-EN

1

Sample graphic for the schematic layout of the operating menu

Additional information regarding:• The arrangement of the parameters according to the menu structure of the
Operation menu, Setup menu, Diagnostics menu with a brief description:
Operating Instructions• Operating concept of the operating menus: Operating Instructions

Endress+Hauser

5About this document

Proline Teqwave M 300 HART

1.3.2

Structure of a parameter description

The individual parts of a parameter description are described in the following section:
Complete parameter nameWrite-protected parameter =Navigation

Navigation path to the parameter via the local display or Web browser
Navigation path to the parameter via the operating tool
The names of the menus, submenus and parameters are abbreviated to the form in which they appear on
the display and in the operating tool.

Prerequisite

The parameter is only available under these specific conditions

Description

Description of the parameter function

Options

List of the individual options for the parameter• Option 1• Option 2

User entry

Parameter entry range

Display

Display value/data of the parameter

Factory setting

Default setting ex works

Additional information

Additional explanations (e.g. in examples):• On individual options• On display values/data• On the input range• On the factory setting• On the parameter function

1.4

Symbols used

1.4.1

Symbols for certain types of information
SymbolMeaning
Tip
Indicates additional information.
Reference to documentation
Reference to pageA

Reference to graphic
Operation via local display
A0028662

Operation via operating tool
A0028663

Write-protected parameterA0028665

6

Endress+Hauser

Proline Teqwave M 300 HART

About this document

1.4.2

Symbols in graphics
SymbolMeaning

1, 2, 3 ...

Item numbers

A, B, C, ...

Views

A-A, B-B, C-C, ...

Sections

1.5

Documentation

1.5.1

Standard documentation

Technical information
DeviceDocumentation code

Proline Teqwave MW 300

TI01763D

Operating instructions
DeviceDocumentation code

Proline Teqwave MW 300 HART

BA02320D

1.5.2

Supplementary device-dependent documentation

Special documentation

Endress+Hauser

Contents

Documentation code

Heartbeat Verification application package (HART)

SD03168D

7

Overview of the Expert operating menu

2

Proline Teqwave M 300 HART

Overview of the Expert operating menu

The following table provides an overview of the menu structure of the expert operating
menu and its parameters. The page reference indicates where the associated description of
the submenu or parameter can be found.

 Expert
Locking status→  11

User role→  12Enter access code→  13‣ System→  13‣ Display→  13‣ Configuration backup→  32‣ Diagnostic handling→  35‣ Administration→  41‣ Sensor→  45‣ Measured values→  46‣ System units→  54‣ Process parameters→  57‣ External process variables→  61‣ Sensor adjustment→  62‣ Factory adjustment→  65‣ I/O configuration8→  66
I/O module 1 to n terminal numbers→  66

I/O module 1 to n information→  66

I/O module 1 to n type→  67

Apply I/O configuration→  67

I/O alteration code→  68Endress+Hauser

Proline Teqwave M 300 HART

Overview of the Expert operating menu

‣ Input→  68‣ Current input 1 to n

→  68‣ Status input 1 to n

→  71‣ Output→  73‣ Current output 1 to n

→  74‣ Pulse/frequency/switch output→  85‣ Relay output 1 to n

→  103

1 to n‣ Communication→  109‣ HART input→  110‣ HART output→  115‣ Web server→  135‣ WLAN settings→  139‣ Application→  146
Reset all totalizers→  146‣ Totalizer 1→  146‣ Diagnostics

Endress+Hauser→  150
Actual diagnostics→  151

Timestamp→  151

Previous diagnostics→  152

Timestamp→  152

Operating time from restart→  152

Operating time→  153‣ Diagnostic list

→  153‣ Device information→  1579

Overview of the Expert operating menu

Proline Teqwave M 300 HART

‣ Main electronic module + I/O→  161‣ Sensor electronic module (ISEM)→  162‣ I/O module 2→  163‣ I/O module 3→  164‣ Display module→  165‣ Data logging→  166‣ Min/max values→  174‣ Heartbeat Technology→  178‣ Simulation→  179

module 1

10Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

3

Description of device parameters

In the following section, the parameters are listed according to the menu structure of thelocal display. Specific parameters for the operating tools are included at the appropriate
points in the menu structure.
Navigation

 Expert

 Expert
Locking status→  11

User role→  12Enter access code→  13‣ System→  13‣ Sensor→  45‣ I/O configuration→  66‣ Input→  68‣ Output→  73‣ Communication→  109‣ Application→  146‣ Diagnostics→  150

Locking status
Navigation

 Expert → Locking status

Description

Displays the active write protection.

User interface• Hardware locked• Temporarily locked

Endress+Hauser

11Description of device parameters

Additional information

Proline Teqwave M 300 HART

Display
If two or more types of write protection are active, the write protection with the highest
priority is shown on the local display. In the operating tool all active types of write
protection are displayed.
Detailed information on access authorization is provided in the "User roles andassociated access authorization" and "Operating concept" sections of the Operations
Instructions for the device
Options
Options

Description

None

The access authorization displayed in the Access status parameter (→  12)
applies . Only appears on local display.

Hardware locked(priority 1)

The DIP switch for hardware locking is activated on the PCB board. This locks write
access to the parameters (e.g. via local display or operating tool) .

Temporarily locked

Write access to the parameters is temporarily locked on account of internal
processes running in the device (e.g. data upload/download, reset, etc.). Once theinternal processing has been completed, the parameters can be changed once
again.

User role
Navigation

 Expert → User role

Description

Displays the access authorization to the parameters via the local display, Web browser or
operating tool.

User interface• Maintenance• Service

Factory setting

Maintenance

Additional information

Description
Access authorization can be modified via the Enter access code parameter(→  13).
If additional write protection is active, this restricts the current access authorizationeven further.
User interface
Detailed information on access authorization is provided in the "User roles andassociated access authorization" and "Operating concept" sections of the Operations
Instructions for the device

12

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Enter access code
Navigation

 Expert → Ent. access code

Description

Use this function to enter the user-specific release code to remove parameter write
protection.

User entry

Max. 16-digit character string comprising numbers, letters and special characters

3.1
Navigation"System" submenu
 Expert → System‣ System

3.1.1
Navigation‣ Display→  13‣ Configuration backup→  32‣ Diagnostic handling→  35‣ Administration→  41"Display" submenu
 Expert → System → Display‣ Display

Endress+Hauser

Format display→  14

Value 1 display→  170% bargraph value 1→  17

100% bargraph value 1→  18

Decimal places 1→  18

Value 2 display→  19Decimal places 2→  19

Value 3 display→  2013

Description of device parameters

Proline Teqwave M 300 HART

0% bargraph value 3→  20100% bargraph value 3→  21

Decimal places 3→  21

Value 4 display→  22

Decimal places 4→  22

Display language→  23

Display interval→  29Display damping→  29Header→  30

Header text

→  30

Separator→  31

Backlight→  32Format display
Navigation

 Expert → System → Display → Format display

Prerequisite

A local display is provided.

Description

Use this function to select how the measured value is shown on the local display.

Selection• 1 value, max. size
• 1 bargraph + 1 value• 2 values• 1 value large + 2 values• 4 values

Factory setting

1 value, max. size

14

Endress+Hauser

Proline Teqwave M 300 HART

Additional information

Description of device parameters

Description
The display format (size, bar graph) and number of simultaneously displayed measured
values (1 to 8) can be configured. This setting only applies to normal operation.
• The Value 1 display parameter (→  17)...Value 8 display parameter(→  28) are used to specify which measured values are shown on the local
display and in what order.
• If more measured values are specified than the display mode selected permits, thenthe values alternate on the device display. The display time until the next change isconfigured using the Display interval parameter (→  29).

Endress+Hauser

15Description of device parameters

Proline Teqwave M 300 HART

Possible measured values shown on the local display:"1 value, max. size" option
XXXXXXXXX

1.00%TS
A0054319"1 bargraph + 1 value" option
XXXXXXXXX
1.00 %TS

10.00
kg/minA0054322"2 values" option
XXXXXXXXX

1.00
%TS

10.00
kg/minA0054323"1 value large + 2 values" option
XXXXXXXXX

1.00%TS
10.00 kg/min
20.0 °C
A0054324"4 values" option
XXXXXXXXX
1.00 %TS
10.00 kg/min
20.0 °C
1000 μS/cm
A0054328

16

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Value 1 display

Navigation

 Expert → System → Display → Value 1 display

Prerequisite• A local display is provided.
• The Load rate option is only available if the volume flow of the medium is read in via theCurrent input 1 to n (→  49) or the fieldbus.

Description

Use this function to select a measured value that is shown on the local display.

Selection• Total solids• Temperature• Electronics temperature• Conductivity• Corrected conductivity• Load rate *• Totalizer 1 *• Current output 1 *• Current output 2 *• Current output 3 *• Current output 4 *Factory setting

Total solids

Additional information

Description
If several measured values are displayed one below the other, the measured value selected
here will be the first value to be displayed. The value is only displayed during normal
operation.
The Format display parameter (→  14) is used to specify how many measured
values are displayed simultaneously and how.
Dependency
The unit of the displayed measured value is taken from the System units submenu(→  54).

0% bargraph value 1Navigation

 Expert → System → Display → 0% bargraph 1

Prerequisite

A local display is provided.

Description

Use this function to enter the 0% bar graph value to be shown on the display for themeasured value 1.

User entry

Signed floating-point number

Factory setting

0 %TS

*

Visibility depends on order options or device settings

Endress+Hauser

17Description of device parameters

Additional information

Proline Teqwave M 300 HART

Description
The Format display parameter (→  14) is used to specify that the measured value
is to be displayed as a bar graph.
User entry
The unit of the displayed measured value is taken from the System units submenu(→  54).

100% bargraph value 1Navigation

 Expert → System → Display → 100% bargraph 1

Prerequisite

A local display is provided.

Description

Use this function to enter the 100% bar graph value to be shown on the display for themeasured value 1.

User entry

Signed floating-point number

Factory setting

Depends on country and nominal diameter

Additional information

Description
The Format display parameter (→  14) is used to specify that the measured value
is to be displayed as a bar graph.
User entry
The unit of the displayed measured value is taken from the System units submenu(→  54).

Decimal places 1Navigation

 Expert → System → Display → Decimal places 1

Prerequisite

A measured value is specified in the Value 1 display parameter (→  17).

Description

Use this function to select the number of decimal places for measured value 1.

Selection•x• x.x• x.xx• x.xxx• x.xxxx

Factory setting

x.xx

Additional information

Description
This setting does not affect the accuracy of the device for measuring or calculating thevalue.

18

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Value 2 display

Navigation

 Expert → System → Display → Value 2 display

Prerequisite• A local display is provided.
• The Load rate option is only available if the volume flow of the medium is read in via theCurrent input 1 to n (→  49) or the fieldbus.

Description

Use this function to select a measured value that is shown on the local display.

Selection• None• Total solids• Temperature• Electronics temperature• Conductivity• Corrected conductivity• Load rate *• Totalizer 1 *• Current output 1 *• Current output 2 *• Current output 3 *• Current output 4 *Factory setting

None

Additional information

Description
If several measured values are displayed one below the other, the measured value selected
here will be the second value to be displayed. The value is only displayed during normal
operation.
The Format display parameter (→  14) is used to specify how many measured
values are displayed simultaneously and how.
Dependency
The unit of the displayed measured value is taken from the System units submenu(→  54).

Decimal places 2Navigation

 Expert → System → Display → Decimal places 2

Prerequisite

A measured value is specified in the Value 2 display parameter (→  19).

Description

Use this function to select the number of decimal places for measured value 2.

Selection•x• x.x• x.xx• x.xxx• x.xxxx*

Visibility depends on order options or device settings

Endress+Hauser

19Description of device parameters

Proline Teqwave M 300 HART

Factory setting

x.xx

Additional information

Description
This setting does not affect the accuracy of the device for measuring or calculating thevalue.

Value 3 display

Navigation

 Expert → System → Display → Value 3 display

Prerequisite• A local display is provided.
• The Load rate option is only available if the volume flow of the medium is read in via theCurrent input 1 to n (→  49) or the fieldbus.

Description

Use this function to select a measured value that is shown on the local display.

Selection

For the picklist, see Value 1 display parameter (→  17)

Factory setting

None

Additional information

Description
If several measured values are displayed one below the other, the measured value selected
here will be the third value to be displayed. The value is only displayed during normal
operation.
The Format display parameter (→  14) is used to specify how many measured
values are displayed simultaneously and how.
Options
The unit of the displayed measured value is taken from the System units submenu(→  54).

0% bargraph value 3Navigation

 Expert → System → Display → 0% bargraph 3

Prerequisite

A selection was made in the Value 3 display parameter (→  20).

Description

Use this function to enter the 0% bar graph value to be shown on the display for themeasured value 3.

User entry

Signed floating-point number

Factory setting

0

20

Endress+Hauser

Proline Teqwave M 300 HART

Additional information

Description of device parameters

Description
The Format display parameter (→  14) is used to specify that the measured value
is to be displayed as a bar graph.
User entry
The unit of the displayed measured value is taken from the System units submenu(→  54).

100% bargraph value 3Navigation

 Expert → System → Display → 100% bargraph 3

Prerequisite

A selection was made in the Value 3 display parameter (→  20).

Description

Use this function to enter the 100% bar graph value to be shown on the display for themeasured value 3.

User entry

Signed floating-point number

Factory setting0

Additional information

Description
The Format display parameter (→  14) is used to specify that the measured value
is to be displayed as a bar graph.
User entry
The unit of the displayed measured value is taken from the System units submenu(→  54).

Decimal places 3Navigation

 Expert → System → Display → Decimal places 3

Prerequisite

A measured value is specified in the Value 3 display parameter (→  20).

Description

Use this function to select the number of decimal places for measured value 3.

Selection•x• x.x• x.xx• x.xxx• x.xxxx

Factory setting

x.xx

Additional information

Description
This setting does not affect the accuracy of the device for measuring or calculating thevalue.

Endress+Hauser

21Description of device parameters

Proline Teqwave M 300 HART

Value 4 display

Navigation

 Expert → System → Display → Value 4 display

Prerequisite• A local display is provided.
• The Load rate option is only available if the volume flow of the medium is read in via theCurrent input 1 to n (→  49) or the fieldbus.

Description

Use this function to select a measured value that is shown on the local display.

Selection

For the picklist, see Value 1 display parameter (→  17)

Factory setting

None

Additional information

Description
If several measured values are displayed one below the other, the measured value selected
here will be the fourth value to be displayed. The value is only displayed during normal
operation.
The Format display parameter (→  14) is used to specify how many measured
values are displayed simultaneously and how.
Options
The unit of the displayed measured value is taken from the System units submenu(→  54).

Decimal places 4Navigation

 Expert → System → Display → Decimal places 4

Prerequisite

A measured value is specified in the Value 4 display parameter (→  22).

Description

Use this function to select the number of decimal places for measured value 4.

Selection•x• x.x• x.xx• x.xxx• x.xxxx

Factory setting

x.xx

Additional information

Description
This setting does not affect the accuracy of the device for measuring or calculating thevalue.

22

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Display language
Navigation

 Expert → System → Display → Display language

Prerequisite

A local display is provided.

Description

Use this function to select the configured language on the local display.

Selection• English• Deutsch• Français• Español• Italiano• Nederlands• Portuguesa• Polski• русский язык (Russian)• Svenska• Türkçe• 中文 (Chinese)• 日本語 (Japanese)• 한국어 (Korean)• čeština (Czech)

Factory setting

English (alternatively, the ordered language is preset in the device)

Value 5 display

Navigation

 Expert → System → Display → Value 5 display

Prerequisite• A local display is provided.
• The Load rate option is only available if the volume flow of the medium is read in via theCurrent input 1 to n (→  49) or the fieldbus.

Description

Use this function to select a measured value that is shown on the local display.

Selection

For the picklist, see Value 1 display parameter (→  17)

Factory setting

None

Additional information

Description
If several measured values are displayed one below the other, the measured value selected
here will be the fifth value to be displayed. The value is only displayed during normal
operation.
The Format display parameter (→  14) is used to specify how many measured
values are displayed simultaneously and how.
Options
The unit of the displayed measured value is taken from the System units submenu(→  54).

Endress+Hauser

23Description of device parameters

Proline Teqwave M 300 HART

0% bargraph value 5Navigation

 Expert → System → Display → 0% bargraph 5

Prerequisite

An option was selected in the Value 5 display parameter (→  23).

Description

Use this function to enter the 0% bar graph value to be shown on the display for themeasured value 5.

User entry

Signed floating-point number

Factory setting0

Additional information

Description
The Format display parameter (→  14) is used to specify that the measured value
is to be displayed as a bar graph.
User entry
The unit of the displayed measured value is taken from the System units submenu(→  54).

100% bargraph value 5Navigation

 Expert → System → Display → 100% bargraph 5

Prerequisite

An option was selected in the Value 5 display parameter (→  23).

Description

Use this function to enter the 100% bar graph value to be shown on the display for themeasured value 5.

User entry

Signed floating-point number

Factory setting0

Additional information

Description
The Format display parameter (→  14) is used to specify that the measured value
is to be displayed as a bar graph.
User entry
The unit of the displayed measured value is taken from the System units submenu(→  54).

Decimal places 5Navigation

 Expert → System → Display → Decimal places 5

Prerequisite

A measured value is specified in the Value 5 display parameter (→  23).

24

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Description

Use this function to select the number of decimal places for measured value 5.

Selection•x• x.x• x.xx• x.xxx• x.xxxx• x.xxxxx• x.xxxxxx

Factory setting

x.xx

Additional information

Description
This setting does not affect the accuracy of the device for measuring or calculating thevalue.

Value 6 display

Navigation

 Expert → System → Display → Value 6 display

Prerequisite• A local display is provided.
• The Load rate option is only available if the volume flow of the medium is read in via theCurrent input 1 to n (→  49) or the fieldbus.

Description

Use this function to select a measured value that is shown on the local display.

Selection

For the picklist, see Value 1 display parameter (→  17)

Factory setting

None

Additional information

Description
If several measured values are displayed one below the other, the measured value selected
here will be the sixth value to be displayed. The value is only displayed during normal
operation.
The Format display parameter (→  14) is used to specify how many measured
values are displayed simultaneously and how.
Options
The unit of the displayed measured value is taken from the System units submenu(→  54).

Decimal places 6Navigation

 Expert → System → Display → Decimal places 6

Prerequisite

A measured value is specified in the Value 6 display parameter (→  25).

Description

Use this function to select the number of decimal places for measured value 6.

Endress+Hauser

25Description of device parameters

Proline Teqwave M 300 HART

Selection•x• x.x• x.xx• x.xxx• x.xxxx• x.xxxxx• x.xxxxxx

Factory setting

x.xx

Additional information

Description
This setting does not affect the accuracy of the device for measuring or calculating thevalue.

Value 7 display

Navigation

 Expert → System → Display → Value 7 display

Prerequisite• A local display is provided.
• The Load rate option is only available if the volume flow of the medium is read in via theCurrent input 1 to n (→  49) or the fieldbus.

Description

Use this function to select a measured value that is shown on the local display.

Selection

For the picklist, see Value 1 display parameter (→  17)

Factory setting

None

Additional information

Description
If several measured values are displayed one below the other, the measured value selected
here will be the seventh value to be displayed. The value is only displayed during normal
operation.
The Format display parameter (→  14) is used to specify how many measured
values are displayed simultaneously and how.
Options
The unit of the displayed measured value is taken from the System units submenu(→  54).

0% bargraph value 7Navigation

 Expert → System → Display → 0% bargraph 7

Prerequisite

An option was selected in the Value 7 display parameter (→  26).

Description

Use this function to enter the 0% bar graph value to be shown on the display for themeasured value 7.

User entry

Signed floating-point number

26

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Factory setting0

Additional information

Description
The Format display parameter (→  14) is used to specify that the measured value
is to be displayed as a bar graph.
User entry
The unit of the displayed measured value is taken from the System units submenu(→  54).

100% bargraph value 7Navigation

 Expert → System → Display → 100% bargraph 7

Prerequisite

An option was selected in the Value 7 display parameter (→  26).

Description

Use this function to enter the 100% bar graph value to be shown on the display for themeasured value 7.

User entry

Signed floating-point number

Factory setting0

Additional information

Description
The Format display parameter (→  14) is used to specify that the measured value
is to be displayed as a bar graph.
User entry
The unit of the displayed measured value is taken from the System units submenu(→  54).

Decimal places 7

Navigation

 Expert → System → Display → Decimal places 7

Prerequisite

A measured value is specified in the Value 7 display parameter (→  26).

Description

Use this function to select the number of decimal places for measured value 7.

Selection•x• x.x• x.xx• x.xxx• x.xxxx• x.xxxxx• x.xxxxxx

Factory setting

x.xx

Endress+Hauser

27Description of device parameters

Additional information

Proline Teqwave M 300 HART

Description
This setting does not affect the accuracy of the device for measuring or calculating thevalue.

Value 8 display

Navigation

 Expert → System → Display → Value 8 display

Prerequisite• A local display is provided.
• The Load rate option is only available if the volume flow of the medium is read in via theCurrent input 1 to n (→  49) or the fieldbus.

Description

Use this function to select a measured value that is shown on the local display.

Selection

For the picklist, see Value 1 display parameter (→  17)

Factory setting

None

Additional information

Description
If several measured values are displayed one below the other, the measured value selected
here will be the eighth value to be displayed. The value is only displayed during normal
operation.
The Format display parameter (→  14) is used to specify how many measured
values are displayed simultaneously and how.
Options
The unit of the displayed measured value is taken from the System units submenu(→  54).

Decimal places 8Navigation

 Expert → System → Display → Decimal places 8

Prerequisite

A measured value is specified in the Value 8 display parameter (→  28).

Description

Use this function to select the number of decimal places for measured value 8.

Selection•x• x.x• x.xx• x.xxx• x.xxxx• x.xxxxx• x.xxxxxx

Factory setting

x.xx

28

Endress+Hauser

Proline Teqwave M 300 HART

Additional information

Description of device parameters

Description
This setting does not affect the accuracy of the device for measuring or calculating thevalue.

Display interval
Navigation

 Expert → System → Display → Display interval

Prerequisite

A local display is provided.

Description

Use this function to enter the length of time the measured values are displayed if thevalues alternate on the display.

User entry

1 to 10 s

Factory setting

5sAdditional information

Description
This type of alternating display only occurs automatically if the number of measured
values defined exceeds the number of values the selected display format can display
simultaneously.
• The Value 1 display parameter (→  17)...Value 8 display parameter (→  28)
are used to specify which measured values are shown on the local display.
• The display format for the measured values displayed is defined in the Format
display parameter (→  14).

Display damping

Navigation

 Expert → System → Display → Display damping

Prerequisite

A local display is provided.

Description

Use this function to enter a time constant for the reaction time of the local display tofluctuations in the measured value caused by process conditions.

User entry

0.0 to 999.9 s

Factory setting

0.0 s

Additional information

User entry
Use this function to enter a time constant (PT1 element 1)) for display damping:• At a low time constant, the display reacts quickly to fluctuating measured variables.
• If a high time constant is entered, the display reacts more slowly.
The damping is not active if the value 0 (factory setting) is entered.

1)

proportional transmission behavior with first order delay

Endress+Hauser

29Description of device parameters

Proline Teqwave M 300 HART

Header

Navigation

 Expert → System → Display → Header

Prerequisite

A local display is provided.

Description

Use this function to select the contents of the header of the local display.

Selection• Device tag• Free text

Factory setting

Device tag

Additional information

Description
The header text only appears during normal operation.
1

XXXXXXXXX

A0029422

1

Position of the header text on the display

Selection• Device tag
Is defined in the Device tag parameter (→  158).
• Free text
Is defined in the Header text parameter (→  30).

Header text

Navigation

 Expert → System → Display → Header text

Prerequisite

The Free text option is selected in the Header parameter (→  30).

Description

Use this function to enter a customer-specific text for the header of the local display.

User entry

Max. 12 characters, such as letters, numbers or special characters (e.g. @, %, /)Factory setting------------

Additional information

Description
The header text only appears during normal operation.

30

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

1

XXXXXXXXX

A0029422

1

Position of the header text on the display

User entry
The number of characters displayed depends on the characters used.

Separator



Navigation

 Expert → System → Display → Separator

Prerequisite

A local display is provided.

Description

Use this function to select the decimal separator.

Selection• . (point)• , (comma)

Factory setting. (point)

Contrast display
Navigation

 Expert → System → Display → Contrast display

Prerequisite

A local display is provided.

Description

Use this function to enter a value to adapt the display contrast to the ambient conditions(e.g. the lighting or viewing angle).

User entry

20 to 80 %Factory setting

Depends on the display

Endress+Hauser

31Description of device parameters

Proline Teqwave M 300 HART

Backlight
Navigation

 Expert → System → Display → Backlight

Prerequisite

One of the following conditions is met:
• Order code for "Display; operation", option F "4-line, illum.; touch control"• Order code for "Display; operation", option G "4-line, illum.; touch control +WLAN"• Order code for "Display; operation", option O "Remote display 4-line illuminated; 10m/
30ft cable; touch control"

Description

Use this function to switch the backlight of the local display on and off.

Selection• Disable• Enable

Factory setting

Enable3.1.2
Navigation"Configuration backup" submenu
 Expert → System → Config. backup‣ Configuration backup
Operating time→  32

Last backup→  33

Configuration management→  33

Backup state→  34Comparison result→  34

Operating time
Navigation

 Expert → System → Config. backup → Operating time

Description

Displays the length of time the device has been in operation.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

Indication
Maximum number of days: 9 999(corresponds to approx. 27 years and 5 months)

32

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Last backup
Navigation

 Expert → System → Config. backup → Last backup

Description

Displays the time since a backup copy of the data was last saved to the device memory.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Configuration management

Navigation

 Expert → System → Config. backup → Config. managem.

Description

Use this function to select an action to save the data to the device memory.

Selection• Cancel• Execute backup• Restore *• Compare *• Clear backup data

Factory setting

CancelAdditional information

Selection
Options

Description

CancelNo action is executed and the user exits the parameter.

Execute backup

A backup copy of the current device configuration is saved from the HistoROM
backup to the memory of the device. The backup copy includes the transmitter dataof the device.
The following message appears on local display: Backup active, please wait!

RestoreThe last backup copy of the device configuration is restored from the device
memory to the device's HistoROM backup. The backup copy includes thetransmitter data of the device.
The following message appears on local display: Restore active! Do not interrupt
power supply!

CompareThe device configuration saved in the device memory is compared with the current
device configuration of the HistoROM backup.
The following message appears on local display: Comparing files
The result can be viewed in Comparison result parameter.

Clear backup data

The backup copy of the device configuration is deleted from the memory of thedevice.
The following message appears on local display: Deleting file

HistoROM
A HistoROM is a "non-volatile" device memory in the form of an EEPROM.

*

Visibility depends on order options or device settings

Endress+Hauser

33

Description of device parameters

Proline Teqwave M 300 HART

Backup state
Navigation

 Expert → System → Config. backup → Backup state

Description

Displays the status of the data backup process.

User interface• None• Backup in progress• Restoring in progress• Delete in progress• Compare in progress• Restoring failed• Backup failed

Factory setting

None

Comparison result
Navigation

 Expert → System → Config. backup → Compar. result

Description

Displays the last result of the comparison of the data records in the device memory and in
the HistoROM.

User interface• Settings identical• Settings not identical• No backup available• Backup settings corrupt• Check not done
• Dataset incompatible

Factory setting

Check not done

Additional information

Description
The comparison is started via the Compare option in the Configuration
management parameter (→  33).
Options

34

Options

Description

Settings identical

The current device configuration of the HistoROM is not identical to the backup
copy in the device memory.
If the transformer configuration of another device has been transmitted to thedevice via HistoROM in Configuration management parameter, the current device
configuration of the HistoROM is only partially identical to the backup copy in thedevice memory: The settings for the transmitter are not identical.

Settings not identical

The current device configuration of the HistoROM is not identical to the backup
copy in the device memory.

No backup available

There is no backup copy of the device configuration of the HistoROM in the device
memory.

Backup settings corrupt

The current device configuration of the HistoROM is corrupt or not compatible withthe backup copy in the device memory.

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Options

Description

Check not done

The device configuration of the HistoROM has not yet been compared to the
backup copy in the device memory.

Dataset incompatible

The backup copy in the device memory is not compatible with the device.

HistoROM
A HistoROM is a "non-volatile" device memory in the form of an EEPROM.

3.1.3
Navigation"Diagnostic handling" submenu
 Expert → System → Diagn. handling‣ Diagnostic handling
Alarm delay→  35‣ Diagnostic behavior→  36

Alarm delay

Navigation

 Expert → System → Diagn. handling → Alarm delay

Description

Use this function to enter the time interval until the device generates a diagnostic
message.
The diagnostic message is reset without a time delay.

User entry

0 to 60 s

Factory setting

0s

Additional information

EffectThis setting affects the following diagnostic messages:• 832 Electronics temperature too high• 833 Electronics temperature too low• 834 Process temperature too high• 835 Process temperature too low• 881 Signal to noise ratio too low• 907 Permittivity out of specification• 908 Volume fraction out of specification• 909 Conductivity out of specification

Endress+Hauser

35Description of device parameters

Proline Teqwave M 300 HART

"Diagnostic behavior" submenu
Each item of diagnostic information is assigned a specific diagnostic behavior at thefactory. The user can change this assignment for specific diagnostic information in the
Diagnostic behavior submenu (→  36).
The following options are available in the Assign behavior of diagnostic no. xxx
parameters:
Options

Description

Alarm

The device stops measurement. The signal outputs and totalizers assume the definedalarm condition. A diagnostic message is generated.
The background lighting changes to red.

Warning

The device continues to measure. The signal outputs and totalizers are not affected. A
diagnostic message is generated.

Logbook entry only

The device continues to measure. The diagnostic message is only displayed in the Eventlogbook submenu (→  155) (Event list submenu (→  156)) and is not displayed inalternating sequence with the operational display.

Off

The diagnostic event is ignored, and no diagnostic message is generated or entered.

For a list of all the diagnostic events, see the Operating Instructions for the device
Navigation

 Expert → System → Diagn. handling → Diagn. behavior‣ Diagnostic behavior

36

Assign behavior of diagnostic no. 441→  37

Assign behavior of diagnostic no. 442→  37

Assign behavior of diagnostic no. 443→  37

Assign behavior of diagnostic no. 444→  38Assign behavior of diagnostic no. 302→  38Assign behavior of diagnostic no. 832→  38Assign behavior of diagnostic no. 833→  39Assign behavior of diagnostic no. 834→  39Assign behavior of diagnostic no. 835→  40Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Assign behavior of diagnostic no. 441 (Current output 1 to n)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no.441

Description

Use this function to change the diagnostic behavior of the 441 Current output 1 to ndiagnostic message.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

Additional informationFor a detailed description of the options available: →  36

Assign behavior of diagnostic no. 442 (Frequency output 1 to n)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no.442

Prerequisite

The measuring device has a pulse/frequency/switch output.

Description

Use this function to change the diagnostic behavior of the 442 Frequency output 1 to ndiagnostic message.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

Additional information

For a detailed description of the options available: →  36

Assign behavior of diagnostic no. 443 (Pulse output 1 to n)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no.443

Prerequisite

The measuring device has a pulse/frequency/switch output.

Description

Use this function to change the diagnostic behavior of the 443 Pulse output 1 to ndiagnostic message.

Selection• Off• Alarm• Warning• Logbook entry only

Endress+Hauser37Description of device parameters

Factory setting
Additional information

Proline Teqwave M 300 HART

Warning
For a detailed description of the options available: →  36

Assign behavior of diagnostic no. 444 (Current input 1 to n)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no.444

Prerequisite

The device has one current input.

Description

Use this function to change the diagnostic behavior of the 444 Current input 1 to ndiagnostic message.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

Additional information

For a detailed description of the options available: →  36

Assign behavior of diagnostic no. 302

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no.302

Description

Use this function to change the diagnostic behavior of the 302 Device verification active
diagnostic message.

Selection• Off• Warning• Logbook entry only

Factory setting

Warning

Additional information

For a detailed description of the options available: →  36

Assign behavior of diagnostic no. 832

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no.832

Description

Use this function to change the diagnostic behavior of the 832 Electronics temperature
too high diagnostic message.

38

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

Additional information

For a detailed description of the options available: →  36

Assign behavior of diagnostic no. 833

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no.833

Description

Use this function to change the diagnostic behavior of the 833 Electronics temperature
too low diagnostic message.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

Additional information

For a detailed description of the options available: →  36

Assign behavior of diagnostic no. 834

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no.834

Description

Use this function to change the diagnostic behavior of the 834 Process temperature toohigh diagnostic message.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

Additional information

Endress+Hauser

For a detailed description of the options available: →  36

39

Description of device parameters

Proline Teqwave M 300 HART

Assign behavior of diagnostic no. 835

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no.835

Description

Use this function to change the diagnostic behavior of the 835 Process temperature too
low diagnostic message.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

Additional information

For a detailed description of the options available: →  36

Assign behavior of diagnostic no. 907

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no.907

Description

Use this function to change the diagnostic behavior of the 907 Permittivity out ofspecification diagnostic message.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

Additional information

For a detailed description of the options available: →  36

Assign behavior of diagnostic no. 908

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no.908

Description

Use this function to change the diagnostic behavior of the 908 Volume fraction out ofspecification diagnostic message.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

40

Endress+Hauser

Proline Teqwave M 300 HART

Additional information

Description of device parameters

For a detailed description of the options available: →  36

3.1.4

"Administration" submenu

Navigation

 Expert → System → Administration‣ Administration‣ Define access code→  41‣ Reset access code→  42

Device reset→  43

Transmitter identifier→  44

Activate SW option→  44

Software option overview→  45"Define access code" wizard
The Define access code wizard (→  41) is only available when operating via thelocal display or Web browser.
If operating via the operating tool, the Define access code parameter can be founddirectly in the Administration submenu. There is no Confirm access code parameter
if the device is operated via the operating tool.
Navigation

Expert → System → Administration → Def. access code

‣ Define access code
Define access code→  41

Confirm access code→  42

Define access code

Navigation

 Expert → System → Administration → Def. access code → Def. access code

Description

Use this function to enter a user-specific release code to restrict write-access to theparameters. This protects the device configuration against any inadvertent modifications
via the local display, Web browser, FieldCare or DeviceCare (via CDI-RJ45 service
interface).

Endress+Hauser

41Description of device parameters

Proline Teqwave M 300 HART

User entry

Max. 16-digit character string comprising numbers, letters and special characters

Additional information

Description
The write protection affects all parameters in the document marked with the  symbol.
On the local display, the  symbol in front of a parameter indicates that the parameter iswrite-protected.
The parameters that cannot be write-accessed are grayed out in the Web browser.
Once the access code has been defined, write-protected parameters can only be
modified if the access code is entered in the Enter access code parameter (→  13).
If you lose the access code, please contact your Endress+Hauser sales organization.
User entry
A message is displayed if the access code is not in the input range.
Factory setting
If the factory setting is not changed or 0 is defined as the access code, the parameters are
not write-protected and the device configuration data can be modified. The user is logged
on in the "Maintenance" role.

Confirm access code

Navigation

 Expert → System → Administration → Def. access code → Confirm code

Description

Enter the defined release code a second time to confirm the release code.

User entry

Max. 16-digit character string comprising numbers, letters and special characters"Reset access code" submenu
Navigation

 Expert → System → Administration → Reset acc. code

‣ Reset access code
Operating time→  42

Reset access code→  43Operating time
Navigation

 Expert → System → Administration → Reset acc. code → Operating time

Description

Displays the length of time the device has been in operation.

42

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

Indication
Maximum number of days: 9 999(corresponds to approx. 27 years and 5 months)

Reset access code
Navigation

 Expert → System → Administration → Reset acc. code → Reset acc. code

Description

Use this function to enter a reset code to reset the user-specific access codes to the factory
setting .

User entry

Character string comprising numbers, letters and special characters

Factory setting

0x00

Additional information

Description
For a reset code, contact your Endress+Hauser service organization.
User entry
The reset code can only be entered via:
• Web browser• DeviceCare, FieldCare (via CDI RJ45 interface)• Fieldbus

Additional parameters in the "Administration" submenu

Device reset

Navigation

 Expert → System → Administration → Device reset

Description

Reset the device configuration - either entirely or in part - to a defined state.

Selection• Cancel• To delivery settings• Restart device• Restore S-DAT backup *Factory setting

Cancel*Visibility depends on order options or device settings

Endress+Hauser

43Description of device parameters

Additional information

Proline Teqwave M 300 HART

Selection
Options

Description

CancelNo action is executed and the user exits the parameter.

To delivery settings

Every parameter for which a customer-specific default setting was ordered is resetto the customer-specific value. All other parameters are reset to the factory setting.

Restart device

The restart resets every parameter with data stored in volatile memory (RAM) to
the factory setting (e.g. measured value data). The device configuration remains
unchanged.

Restore S-DAT backup

Restores the data that is saved on the S-DAT. Additional information: This function
can be used to resolve the memory issue "083 Memory content inconsistent" or to
restore the S-DAT data when a new S-DAT has been installed.

 This option is displayed only in an alarm condition.

Transmitter identifierNavigation

 Expert → System → Administration → Transm. identif.

Description

Select transmitter identifier.

User interface• Unknown
• 500
• 300

Factory setting

300

Activate SW option



Navigation

 Expert → System → Administration → Activate SW opt.

Description

Use this function to enter an activation code to enable an additional, ordered software
option.

User entry

Max. 10-digit string of numbers.

Factory setting

Depends on the software option ordered

Additional information

Description
If a measuring device was ordered with an additional software option, the activation codeis programmed in the device at the factory.
To activate a software option subsequently, please contact your Endress+Hauser salesorganization.
Entering the activation code
The activation code is linked to the serial number of the measuring device and varies
according to the device and software option.

44

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

If an incorrect or invalid code is entered, this results in the loss of software options thathave already been activated.
▸ Before you enter a new activation code, make a note of the current activation code .
▸ Enter the new activation code provided by Endress+Hauser when the new software
option was ordered.
▸ If the code entered is incorrect or invalid, enter the old activation code .
▸ Have your Endress+Hauser sales organization check the new activation code
remembering to specify the serial number or ask for the code again.
Example for a software option
Order code for "Application package", option EA "Extended HistoROM"
Web browser
Once a software option has been activated, the page must be loaded again in the Webbrowser.

Software option overview
Navigation

 Expert → System → Administration → SW option overv.

Description

Displays all the software options that are enabled in the device.

User interface
Additional information

Description
Displays all the options that are available if ordered by the customer.
"Extended HistoROM" option
Order code for "Application package", option EA "Extended HistoROM"

3.2
Navigation"Sensor" submenu
 Expert → Sensor‣ Sensor

Endress+Hauser‣ Measured values→  46‣ System units→  54‣ Process parameters→  57‣ External process variables→  6145

Description of device parameters

Proline Teqwave M 300 HART

3.2.1

‣ Sensor adjustment→  62‣ Factory adjustment→  65"Measured values" submenu

Navigation

 Expert → Sensor → Measured val.

‣ Measured values‣ Process variables→  46‣ Totalizer→  48‣ Input values→  49‣ Output values→  50"Process variables" submenu
Navigation

 Expert → Sensor → Measured val. → Process variab.

‣ Process variables
Total solids→  46

Temperature→  47

Electronics temperature→  47

Conductivity→  47

Corrected conductivity→  47

Load rate→  47

Total solids
Navigation

 Expert → Sensor → Measured val. → Process variab. → Total solids

Description

Shows total solids (fraction of total weight or concentration per volume unit).

User interface

Signed floating-point number

46

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Temperature
Navigation

 Expert → Sensor → Measured val. → Process variab. → Temperature

Description

Shows the medium temperature currently measured.

User interface

Signed floating-point number

Electronics temperature
Navigation

 Expert → Sensor → Measured val. → Process variab. → Electronics temp

Description

Shows the electronics temperature currently measured.

User interface

Signed floating-point number

Conductivity
Navigation

 Expert → Sensor → Measured val. → Process variab. → Conductivity

Description

Shows the conductivity currently measured.

User interface

Floating-point number

Corrected conductivity
Navigation

 Expert → Sensor → Measured val. → Process variab. → CorrConductivity

Description

Shows the conductivity measured compensated for temperature.

User interface

Floating-point number

Load rate
Navigation

 Expert → Sensor → Measured val. → Process variab. → Load rate

Prerequisite

The volume flow of the medium is read in via the Current input 1 to n (→  49) or thefieldbus.

Description

Shows the total solids flow rate.

Endress+Hauser

47

Description of device parameters

User interface

Proline Teqwave M 300 HART

Signed floating-point number"Totalizer" submenu
Navigation

 Expert → Sensor → Measured val. → Totalizer‣ Totalizer
Totalizer 1 value→  48

Totalizer 1 overflow→  48

Totalizer 1 value
Navigation

 Expert → Sensor → Measured val. → Totalizer → Tot. 1 value

Description

Displays the current totalizer reading.

User interface

Signed floating-point number

Additional information

Description
As it is only possible to display a maximum of 7 digits in the operating tool, the current
counter value is the sum of the totalizer value and the overflow value from the Totalizer
overflow 1 to n parameter if the display range is exceeded.
In the event of an error, the totalizer adopts the mode defined in the Failure modeparameter (→  149).
Display
The unit of the selected process variable is defined in the Unit totalizer parameter(→  147) for the totalizer.

Totalizer 1 overflow

Navigation

 Expert → Sensor → Measured val. → Totalizer → Tot. 1 overflow

Description

Displays the current totalizer overflow.

User interface

Integer with sign

Additional information

Description
If the current totalizer reading exceeds 7 digits, which is the maximum value range thatcan be displayed by the operating tool, the value above this range is output as an overflow.

48

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

The current totalizer value is therefore the sum of the overflow value and the totalizer
value from the Totalizer value 1 to n parameter.
Display
The unit of the selected process variable is defined in the Unit totalizer parameter(→  147) for the totalizer.

"Input values" submenu
Navigation

 Expert → Sensor → Measured val. → Input values‣ Input values‣ Current input 1 to n

→  49‣ Value status input 1 to n

→  50"Current input 1 to n" submenu
Navigation

 Expert → Sensor → Measured val. → Input values → Current
input 1 to n

‣ Current input 1 to n
Measured values 1 to n

→  49Measured current 1 to n

→  50Measured values 1 to n
Navigation

 Expert → Sensor → Measured val. → Input values → Current input 1 to n
→ Measured val. 1 to n

Description

Displays the current input value.

User interface

Signed floating-point number

Endress+Hauser

49

Description of device parameters

Proline Teqwave M 300 HART

Measured current 1 to n
Navigation

 Expert → Sensor → Measured val. → Input values → Current input 1 to n → Measur.
curr. 1 to n

Description

Displays the current value of the current input.

User interface

0 to 22.5 mA"Value status input 1 to n" submenu
Navigation

 Expert → Sensor → Measured val. → Input values → Val.stat.inp.
1 to n‣ Value status input 1 to n
→  50

Value status input

Value status input
Navigation

 Expert → Sensor → Measured val. → Input values → Val.stat.inp. 1 to n
→ Val.stat.inp.

Description

Displays the current input signal level.

User interface• High• Low"Output values" submenu
Navigation

 Expert → Sensor → Measured val. → Output values‣ Output values‣ Value current output 1 to n

→  51‣ Pulse/frequency/switch output→  51‣ Relay output 1 to n

→  53

1 to n50

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters"Value current output 1 to n" submenu
Navigation

 Expert → Sensor → Measured val. → Output values → Current
output 1 to n

‣ Value current output 1 to n
Output current→  51

Measured current→  51

Output current
Navigation

 Expert → Sensor → Measured val. → Output values → Current output 1 to n
→ Output curr.

Description

Displays the current value currently calculated for the current output.

User interface

0 to 22.5 mA

Measured current
Navigation

 Expert → Sensor → Measured val. → Output values → Current output 1 to n
→ Measur. curr.

Description

Displays the actual measured value of the output current.

User interface

0 to 30 mA"Pulse/frequency/switch output 1 to n" submenu
Navigation

 Expert → Sensor → Measured val. → Output values → PFS output
1 to n‣ Pulse/frequency/switch output
1 to nEndress+Hauser

Output frequency→  52

Pulse output→  52

Switch state→  5351

Description of device parameters

Proline Teqwave M 300 HART

Output frequency
Navigation

 Expert → Sensor → Measured val. → Output values → PFS output 1 to n → Output
freq.

Prerequisite

In the Operating mode parameter (→  87), the Frequency option is selected.

Description

Displays the actual value of the output frequency which is currently measured.

User interface

0.0 to 12 500.0 Hz

Pulse output
Navigation

 Expert → Sensor → Measured val. → Output values → PFS output 1 to n → Pulse
outputPrerequisite

The Pulse option is selected in the Operating mode parameter (→  87) parameter.

Description

Displays the pulse frequency currently output.

User interface

Positive floating-point number

Additional information

Description• The pulse output is an open collector output.
• This is configured at the factory in such a way that the transistor is conductive for theduration of the pulse (NO contact) and is safety-oriented.

1

1

t

0
NC

t

0
NO

A0028726

0
1
NC
NO

Non-conductive
Conductive
NC contact (normally closed)
NO contact (normally open)

The output behavior can be reversed via the Invert output signal parameter (→  103)
i.e. the transistor does not conduct for the duration of the pulse.
In addition, the behavior of the output in the event of a device alarm (Failure modeparameter (→  91)) can be configured.

52

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Switch state
Navigation

 Expert → Sensor → Measured val. → Output values → PFS output 1 to n → Switch
state

Prerequisite

The Switch option is selected in the Operating mode parameter (→  87).

Description

Displays the current switch status of the status output.

User interface• Open• Closed

Additional information

User interface• OpenThe switch output is not conductive.
• Closed
The switch output is conductive.

"Relay output 1 to n" submenu
Navigation

 Expert → Sensor → Measured val. → Output values → Relay
output 1 to n

‣ Relay output 1 to n
Switch state→  53

Switch cycles→  54Max. switch cycles number→  54

Switch state
Navigation

 Expert → Sensor → Measured val. → Output values → Relay output 1 to n → Switch
state

Description

Displays the current status of the relay output.

User interface• Open• Closed

Additional information

User interface• OpenThe relay output is not conductive.
• Closed
The relay output is conductive.

Endress+Hauser

53Description of device parameters

Proline Teqwave M 300 HART

Switch cycles
Navigation

 Expert → Sensor → Measured val. → Output values → Relay output 1 to n → Switch
cyclesDescription

Displays all the switch cycles performed.

User interface

Positive integer

Max. switch cycles number
Navigation

 Expert → Sensor → Measured val. → Output values → Relay output 1 to n → Max.
cycles no.

Description

Displays the maximum number of guaranteed switch cycles.

User interface

Positive integer

3.2.2"System units" submenu

Navigation

 Expert → Sensor → System units‣ System units

54

Total solids unit

→  55

Density unit

→  55Mass flow unit

→  55Mass unit→  56

Volume flow unit

→  56

Temperature unit

→  56

Conductivity unit

→  57

Date/time format→  57

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Total solids unit
Navigation

 Expert → Sensor → System units → TotalSolidsUnit

Description

Select total solids unit.

Selection

SI units
• %TS
• ppm
• g/l
• mg/l• kg/m³• mg/cm³

Factory setting

Depends on country

US units• lb/gal (us)• lb/ft³

Density unitNavigation

 Expert → Sensor → System units → Density unit

Description

Select density unit.

Selection

SI units• g/l• mg/l• kg/m³• mg/cm³

Factory setting

Depends on country

US units• lb/gal (us)• lb/ft³

Mass flow unitNavigation

 Expert → Sensor → System units → Mass flow unit

Prerequisite

The volume flow of the medium is read in via the Current input 1 to n (→  49).

Description

Select mass flow unit.

Selection

SI units• kg/h• kg/d• t/h• t/dFactory setting

Depends on country

Endress+Hauser

US units• oz/h• oz/d• lb/h• lb/d• STon/h• STon/d

55

Description of device parameters

Proline Teqwave M 300 HART

Mass unitNavigation

 Expert → Sensor → System units → Mass unit

Prerequisite

The volume flow of the medium is read in via the Current input 1 to n (→  49) or thefieldbus.

Description

Select mass unit.

Selection

SI units• kg•t

Factory setting

Depends on country

US units• oz• lb• STonVolume flow unitNavigation

 Expert → Sensor → System units → Volume flow unit

Prerequisite

The volume flow of the medium is read in via the Current input 1 to n (→  49).

Description

Select volume flow unit.

Selection

SI units
• l/h
• l/s
• dm³/min• m³/h• m³/s

Factory settingl/hTemperature unitNavigation

 Expert → Sensor → System units → Temperature unit

Description

Select temperature unit.

Selection

SI units• °C•KFactory setting

Depends on country

56

US units• °F• °R

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Conductivity unitNavigation

 Expert → Sensor → System units → Conductiv. unit

Description

Select conductivity unit.

Selection

SI units• nS/cm• µS/cm• µS/m• µS/mm• mS/m• mS/cm• S/cm• S/mFactory settingµS/cm

Date/time format



Navigation

 Expert → Sensor → System units → Date/time format

Description

Select date and time format.

Selection• dd.mm.yy hh:mm• dd.mm.yy hh:mm am/pm• mm/dd/yy hh:mm• mm/dd/yy hh:mm am/pm

Factory setting

dd.mm.yy hh:mm

3.2.3
Navigation"Process parameters" submenu
 Expert → Sensor → Process param.

‣ Process parameters

Endress+Hauser

Temperature damping→  58

Conductivity damping→  58

Total solids override→  58‣ Total solids monitoring→  58‣ Partially filled pipe detection→  6057

Description of device parameters

Proline Teqwave M 300 HART

Temperature damping

Navigation

 Expert → Sensor → Process param. → Temp. damping

Description

Enter a time constant for damping (PT1 element) of the temperature measured value.
Damping reduces the effect of measured value fluctuations.

User entry

0 to 999.9 s

Factory setting

0s

Conductivity damping

Navigation

 Expert → Sensor → Process param. → Conduct. damping

Description

Enter a time constant for damping (PT1 element) of the conductivity measured value.
Damping reduces the effect of measured value fluctuations.

User entry

0 to 999.9 s

Factory setting

0s

Total solids override

Navigation

 Expert → Sensor → Process param. → TotalSolOverride

Description

If suppression of the total solids measurement is enabled (On option), zero is output forthe measured value. This is suitable for the cleaning processes for the pipeline, forexample.

Selection

• Off
• On

Factory setting

Off"Total solids monitoring" submenu
Navigation

 Expert → Sensor → Process param. → TotSolidsMonitor‣ Total solids monitoring
Assign process variable58→  59Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Lower range limit→  59Upper range limit→  59

Response time→  60Assign process variable

Navigation

 Expert → Sensor → Process param. → TotSolidsMonitor → Assign variable

Description

Select the process variable for total solids monitoring.

Selection• Off• Total solids

Factory setting

Total solids

Lower range limitNavigation

 Expert → Sensor → Process param. → TotSolidsMonitor → LowerRangeLimit

Description

Enter the lower limit value for the measuring range of the total solids.

User entry

Signed floating-point number

Factory setting–1 %TS

Upper range limitNavigation

 Expert → Sensor → Process param. → TotSolidsMonitor → UpperRangeLimit

Description

Enter the upper limit value for the measuring range of the total solids.

User entry

Signed floating-point number

Factory setting

51 %TS

Endress+Hauser

59Description of device parameters

Proline Teqwave M 300 HART

Response timeNavigation

 Expert → Sensor → Process param. → TotSolidsMonitor → Response time

Description

Enter a delay until the diagnostic message is generated in the event the measuring range
is exceeded.

User entry

0 to 100 s

Factory setting

60 s"Partially filled pipe detection" submenu
Navigation

 Expert → Sensor → Process param. → Partial pipe det‣ Partially filled pipe detection
Partially filled pipe detection→  60

Response time→  61

Partially filled pipe detection
Navigation

 Expert → Sensor → Process param. → Partial pipe det → Partial pipe det

Description

If activated, a diagnostic message is generated if the antennas are no longer in full contact
with the medium.

Selection

• Off
• On

Factory setting

Off

Threshold

Navigation

 Expert → Sensor → Process param. → Partial pipe det → Threshold

Description

Enter threshold for partially filled pipe detection. If the measured value drops below thethreshold, a diagnostic message is generated.

User entry

Decibel as negative floating point number

Factory setting

–6 dB

60

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Response time
Navigation

 Expert → Sensor → Process param. → Partial pipe det → Response time

Description

Enter a delay until the diagnostic message is generated in the event the pipe is detected as
partially filled.

User entry

0 to 20.0 s

Factory setting

3s

3.2.4

"External compensation" submenu

Navigation

 Expert → Sensor → External comp.

‣ External process variables
Volume flow source→  61

Volume flow

→  61

Volume flow source

Navigation

 Expert → Sensor → ExternalProcVar. → VolumeFlowSource

Description

Select the input via which the measured value of the volume flow is read in. The volume
flow is used to calculate the load rate.

Selection• Off• Current input 1 *• Current input 2 *• Current input 3 *• External value

Factory setting

Off

Volume flow
Navigation

 Expert → Sensor → ExternalProcVar. → Volume flow

Description

Shows the volume flow reported by the external measuring device.

*

Visibility depends on order options or device settings

Endress+Hauser

61Description of device parameters

User entry

Proline Teqwave M 300 HART

Signed floating-point number

3.2.5"Sensor adjustment" submenu

Navigation

 Expert → Sensor → Sensor adjustm.

‣ Sensor adjustment
Measuring interval→  62‣ Process variable adjustment→  62

Measuring interval

Navigation

 Expert → Sensor → Sensor adjustm. → Measur. interval

Description

Displays the interval between two measuring periods.

User interface

0 to 10 000 ms"Process variable adjustment" submenu
Navigation

 Expert → Sensor → Sensor adjustm. → Variable adjust‣ Process variable adjustment

62

Temperature offset→  63Temperature factor→  63

Conductivity offset→  63

Conductivity factor→  63

Corrected conductivity offset→  64

Corrected conductivity factor→  64

Electronics temperature offset→  64

Electronics temperature factor→  64Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Load rate offset→  65

Load rate factor→  65Temperature offset

Navigation

 Expert → Sensor → Sensor adjustm. → Variable adjust → Temp. offset

Description

Enter the offset by which to shift the zero point for temperature.

User entry

Signed floating-point number

Factory setting0

Temperature factor



Navigation

 Expert → Sensor → Sensor adjustm. → Variable adjust → Temp. factor

Description

Enter the multiplication factor to apply to the temperature value.

User entry

Positive floating-point number

Factory setting

1

Conductivity offset

Navigation

 Expert → Sensor → Sensor adjustm. → Variable adjust → Conduct. offset

Description

Enter the offset by which to shift the zero point for conductivity.

User entry

Signed floating-point number

Factory setting0

Conductivity factor



Navigation

 Expert → Sensor → Sensor adjustm. → Variable adjust → Conduct. factor

Description

Enter the multiplication factor to apply to the conductivity value.

User entry

Positive floating-point number

Endress+Hauser

63Description of device parameters

Factory setting

Proline Teqwave M 300 HART

1

Corrected conductivity offset

Navigation

 Expert → Sensor → Sensor adjustm. → Variable adjust → Corr.cond.offset

Description

Enter the offset by which to shift the zero point for the corrected conductivity.

User entry

Signed floating-point number

Factory setting0

Corrected conductivity factor



Navigation

 Expert → Sensor → Sensor adjustm. → Variable adjust → Corr.cond.factor

Description

Enter the multiplication factor to apply to the corrected conductivity value.

User entry

Positive floating-point number

Factory setting

1

Electronics temperature offset

Navigation

 Expert → Sensor → Sensor adjustm. → Variable adjust → ElectrTempOffset

Description

Enter the offset by which to shift the zero point for the electronics temperature.

User entry

Signed floating-point number

Factory setting0Electronics temperature factor



Navigation

 Expert → Sensor → Sensor adjustm. → Variable adjust → ElectrTempFactor

Description

Enter the multiplication factor to apply to the electronics temperature.

User entry

Positive floating-point number

Factory setting

1

64

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Load rate offset

Navigation

 Expert → Sensor → Sensor adjustm. → Variable adjust → Load rate offset

Prerequisite

The volume flow of the medium is read in via the Current input 1 to n (→  49).

Description

Enter the offset by which to shift the zero point for the load rate.

User entry

Signed floating-point number

Factory setting0

Load rate factor



Navigation

 Expert → Sensor → Sensor adjustm. → Variable adjust → Load rate factor

Prerequisite

The volume flow of the medium is read in via the Current input 1 to n (→  49).

Description

Enter the multiplication factor to apply to the load rate value.

User entry

Positive floating-point number

Factory setting

1

3.2.6
Navigation"Factory adjustment" submenu
 Expert → Sensor → FactoryAdjustm.

‣ Factory adjustment
Nominal diameter→  65Date/time→  66

Nominal diameter
Navigation

 Expert → Sensor → FactoryAdjustm. → Nominal diameter

Description

Shows the nominal diameter of the sensor.

User interface

Character string comprising numbers, letters and special characters

Endress+Hauser

65Description of device parameters

Proline Teqwave M 300 HART

Date/timeNavigation

 Expert → Sensor → FactoryAdjustm. → Date/time

Description

Shows the date and time of the factory adjustment.

User interface

Character string comprising numbers, letters and special characters

3.3"I/O configuration" submenu

Navigation

 Expert → I/O config.

‣ I/O configuration
I/O module 1 to n terminal numbers→  66

I/O module 1 to n information→  66

I/O module 1 to n type→  67

Apply I/O configuration→  67

I/O alteration code→  68I/O module 1 to n terminal numbers
Navigation

 Expert → I/O config. → I/O 1 to n terminals

Description

Displays the terminal numbers used by the I/O module.

User interface• Not used
• 26-27 (I/O 1)• 24-25 (I/O 2)• 22-23 (I/O 3)

I/O module 1 to n information
Navigation

 Expert → I/O config. → I/O 1 to n info

Description

Displays information about the plugged in I/O module.

66

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

User interface• Not plugged• Invalid• Not configurable• Configurable• HART

Additional information"Not plugged" option
The I/O module is not plugged in.
"Invalid" option
The I/O module is not plugged correctly.
"Not configurable" option
The I/O module is not configurable.
"Configurable" option
The I/O module is configurable.
"Fieldbus" option
The I/O module is configured for HART.

I/O module 1 to n type

Navigation

 Expert → I/O config. → I/O 1 to n type

Prerequisite

For the following order code:• "Output; input 2", option D "Configurable I/O initial setting off"• "Output; input 3", option D "Configurable I/O initial setting off"

Description

Use this function to select the I/O module type for the configuration of the I/O module.

Selection• Off• Current output *• Current input *• Status input *• Pulse/frequency/switch output *• Relay output *Factory setting

Off

Apply I/O configuration



Navigation

 Expert → I/O config. → Apply I/O config

Description

Use this function to activate the newly configured I/O module type.

*

Visibility depends on order options or device settings

Endress+Hauser

67Description of device parameters

Proline Teqwave M 300 HART

Selection

• No
• Yes

Factory settingNo

I/O alteration code

Navigation

 Expert → I/O config. → I/O alterat.code

Description

Use this function to enter the ordered activation code to activate the I/O configuration
change.

User entry

Positive integer

Factory setting0

Additional information

Description
The I/O configuration is changed in the I/O module type parameter (→  67).

3.4
Navigation"Input" submenu
 Expert → Input‣ Input

3.4.1
Navigation‣ Current input 1 to n

→  68‣ Status input 1 to n

→  71"Current input 1 to n" submenu
 Expert → Input → Current input 1 to n

‣ Current input 1 to n

68

Terminal number→  69Signal mode→  69Current span→  690/4 mA value→  7020 mA value→  70Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Failure mode→  71

Failure value→  71

Terminal number
Navigation

 Expert → Input → Current input 1 to n → Terminal no.

Description

Displays the terminal numbers used by the current input module.

User interface• Not used
• 24-25 (I/O 2)• 22-23 (I/O 3)

Additional information"Not used" option
The current input module does not use any terminal numbers.

Signal modeNavigation

 Expert → Input → Current input 1 to n → Signal mode

Description

Use this function to select the signal mode for the current input.

Selection• Passive• Active *Factory setting

Passive

Current spanNavigation

 Expert → Input → Current input 1 to n → Current span

Description

Use this function to select the current range for the process value output and the upperand lower level for signal on alarm.

Selection• 4...20 mA (4...20.5 mA)• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 0...20 mA (0...20.5 mA)*

Visibility depends on order options or device settings

Endress+Hauser

69Description of device parameters

Proline Teqwave M 300 HART

Factory setting

Country-specific:• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)

Additional information

ExamplesSample values for the current range: Current span parameter (→  75)

0/4 mA value

Navigation

 Expert → Input → Current input 1 to n → 0/4 mA value

Description

Enter 4 mA value.

User entry

Signed floating-point number

Factory setting

0 % TS

Additional information

Current input behavior
The current input behaves differently depending on the settings configured in thefollowing parameters:• Current span (→  69)• Failure mode (→  71)
Configuration examples
Pay attention to the configuration examples for 4 mA value parameter (→  77).

20 mA value

Navigation

 Expert → Input → Current input 1 to n → 20 mA value

Description

Enter 20 mA value.

User entry

Signed floating-point number

Factory setting

12 %TS

Additional information

Configuration examples
Pay attention to the configuration examples for 4 mA value parameter (→  77).

70

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Failure modeNavigation

 Expert → Input → Current input 1 to n → Failure mode

Description

Use this function to select the input behavior when measuring a current outside theconfigured Current span parameter (→  69).

Selection• Alarm• Last valid value• Defined value

Factory setting

Alarm

Additional information

Options• Alarm
An error message is set.
• Last valid value
The last valid measured value is used.
• Defined value
A user-defined measured value is used (Failure value parameter (→  71)).

Failure value

Navigation

 Expert → Input → Current input 1 to n → Failure value

Prerequisite

In the Failure mode parameter (→  71), the Defined value option is selected.

Description

Use this function to enter the value that the device uses if it does not receive an input
signal from the external device, or if the input signal is invalid.

User entry

Signed floating-point number

Factory setting

0

3.4.2

"Status input 1 to n" submenu

Navigation

 Expert → Input → Status input 1 to n

‣ Status input 1 to n

Endress+Hauser

Terminal number→  72

Assign status input→  72

Value status input→  7371

Description of device parameters

Proline Teqwave M 300 HART

Active level→  73

Response time status input→  73

Terminal number
Navigation

 Expert → Input → Status input 1 to n → Terminal no.

Description

Displays the terminal numbers used by the status input module.

User interface• Not used
• 24-25 (I/O 2)• 22-23 (I/O 3)

Additional information"Not used" option
The status input module does not use any terminal numbers.

Assign status input



Navigation

 Expert → Input → Status input 1 to n → Assign stat.inp.

Description

Use this function to select the function for the status input.

Selection• Off• Reset totalizer 1• Flow override

Factory setting

Off

Additional information

Options• Off
The status input is switched off.
• Reset totalizer 1
The totalizer is reset.
• Flow override
The Flow override is activated.
Note on the Flow override:• The Flow override is enabled as long as the level is at the status input (continuous
signal).
• All other assignments react to a change in level (pulse) at the status input.

72

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Value status input
Navigation

 Expert → Input → Status input 1 to n → Val.stat.inp.

Description

Displays the current input signal level.

User interface• High• Low

Active level

Navigation

 Expert → Input → Status input 1 to n → Active level

Description

Use this function to determine the input signal level at which the assigned function is
activated.

Selection• High• Low

Factory settingHigh

Response time status input



Navigation

 Expert → Input → Status input 1 to n → Response time

Description

Use this function to enter the minimum time period for which the input signal level must
be present before the selected function is activated.

User entry

5 to 200 ms

Factory setting

50 ms

3.5
Navigation"Output" submenu
 Expert → Output‣ Output‣ Current output 1 to n

Endress+Hauser→  7473

Description of device parameters

Proline Teqwave M 300 HART

‣ Pulse/frequency/switch output→  85‣ Relay output 1 to n

→  103

1 to n3.5.1"Current output 1 to n" submenu

Navigation

 Expert → Output → Curr.output 1 to n

‣ Current output 1 to n
Terminal number→  74Signal mode→  75

Process variable current output→  75Current range output→  75

Fixed current→  76Lower range value output→  77Upper range value output→  78

Measuring mode current output→  78Damping current output→  83

Failure behavior current output→  83

Failure current→  84

Output current→  84

Measured current→  85

Terminal number
Navigation

 Expert → Output → Curr.output 1 to n → Terminal no.

Description

Displays the terminal numbers used by the current output module.

User interface• Not used
• 26-27 (I/O 1)• 24-25 (I/O 2)• 22-23 (I/O 3)

74

Endress+Hauser

Proline Teqwave M 300 HART

Additional information

Description of device parameters"Not used" option
The current output module does not use any terminal numbers.

Signal modeNavigation

 Expert → Output → Curr.output 1 to n → Signal mode

Description

Use this function to select the signal mode for the current output.

Selection• Active *• Passive *Factory setting

ActiveProcess variable current output

Navigation

 Expert → Output → Curr.output 1 to n → Proc.var. outp

Prerequisite

The Load rate option is only available if the volume flow of the medium is read in via theCurrent input 1 to n (→  49) or the fieldbus.

Description

Use this function to select a process variable for the current output.

Selection• Off• Total solids• Temperature• Electronics temperature• Conductivity• Corrected conductivity• Load rate *Factory setting

Total solids

Current range output

Navigation

 Expert → Output → Curr.output 1 to n → Curr.range out

Description

Select current range for process value output and upper/lower level for alarm signal.

Selection• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4...20.5 mA)• 0...20 mA (0...20.5 mA)• Fixed value*

Visibility depends on order options or device settings

Endress+Hauser

75Description of device parameters

Proline Teqwave M 300 HART

Factory setting

Depends on country:• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)

Additional information

Description• In the event of a device alarm, the current output adopts the value specified in the
Failure mode parameter (→  83).
• The measuring range is specified via the Lower range value output parameter(→  77) and Upper range value output parameter (→  78).
"Fixed current" option• This option is used for a HART Multidrop network.
• It can only be used for the 4…20 mA HART current output (current output 1).
• The current value is set via the Fixed current parameter (→  76).
ExampleShows the relationship between the current range for the output of the process value andthe two signal on alarm levels:
2

1

3

I [mA]
A0034351

1
2
3

Current range for process value
Lower level for signal on alarm
Upper level for signal on alarm

Selection
Selection

1

2

3

4...20 mA NE (3.8...20.5 mA)

3.8 to 20.5 mA< 3.6 mA

> 21.95 mA

4...20 mA US (3.9...20.8 mA)

3.9 to 20.8 mA US< 3.6 mA

> 21.95 mA

4...20 mA (4...20.5 mA)

4 to 20.5 mA< 3.6 mA

> 21.95 mA

0...20 mA (0...20.5 mA)

0 to 20.5 mA

0 mA> 21.95 mA

Fixed current

Navigation

 Expert → Output → Curr.output 1 to n → Fixed current

Prerequisite

The Fixed current option is selected in the Current span parameter (→  75).

Description

Use this function to enter a constant current value for the current output.

User entry

0 to 22.5 mA

Factory setting

22.5 mA

76

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Lower range value output

Navigation

 Expert → Output → Curr.output 1 to n → Low.range outp

Prerequisite

In Current span parameter (→  75), one of the following options is selected:• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4...20.5 mA)• 0...20 mA (0...20.5 mA)

Description

Use this function to enter a value for the start of measuring range.

User entry

Floating point number with sign

Factory setting

0 %TS

Additional information

Description
Positive and negative values are permitted depending on the process variable assigned in
the Assign current output parameter (→  75). In addition, the value can be greaterthan or smaller than the value assigned for the 20 mA current in the Upper range value
output parameter (→  78).
Dependency
The unit depends on the process variable selected in the Assign current output
parameter (→  75).
Current output behavior
The current output behaves differently depending on the settings configured in thefollowing parameters:• Current span (→  75)• Failure mode (→  83)
Configuration examples
I [mA]
20
84Q-250 01+750
1A0013757

Q
I
1

Endress+Hauser

Flow
Current
Measuring range is exceeded or undershot

77Description of device parameters

Proline Teqwave M 300 HART

Upper range value output

Navigation

 Expert → Output → Curr.output 1 to n → Upp.range outp

Prerequisite

In Current span parameter (→  75), one of the following options is selected:• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4...20.5 mA)• 0...20 mA (0...20.5 mA)

Description

Use this function to enter a value for the end of measuring range.

User entry

Floating point number with sign

Factory setting

12 %TS

Additional information

Description
Positive and negative values are permitted depending on the process variable assigned in
the Assign current output parameter (→  75). In addition, the value can be greaterthan or smaller than the value assigned for the 0/4 mA current in the Lower range value
output parameter (→  77).
Dependency
The unit depends on the process variable selected in the Assign current output
parameter (→  75).
ExampleConfiguration examples
Pay attention to the configuration examples for the Lower range value output
parameter (→  77).

Measuring mode current output

Navigation

 Expert → Output → Curr.output 1 to n → Output mode

Prerequisite

The following option is selected in the Process variable current output parameter(→  75):
Load rate
One of the following options is selected in the Current span parameter (→  75):
• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4...20.5 mA)• 0...20 mA (0...20.5 mA)

Description

78

Use this function to select the measuring mode for the current output.

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Selection• Forward flow
• Forward/Reverse flow *• Reverse flow compensation

Factory setting

Forward flow

Additional information

Description
The process variable that is assigned to the current output via the Assign current
output parameter (→  75) is displayed below the parameter.
"Forward flow" option
The current output signal is proportional to the process variable assigned. The measuring
range is defined by the values that are assigned to the Lower range value output
parameter (→  77) and the Upper range value output parameter (→  78).
The flow components outside the scaled measuring range are taken into account for signal
output as follows:
Both values are defined such that they are not equal to zero flow e.g.:• Start of measuring range = –50 kg/h
• End of measuring range = 100 kg/h
"Forward/Reverse flow" option
I [mA]
20

4
0

Q

0
3

1

2
A0013758

I
Q
1
2
3

Current
Flow
Start of measuring range output (0/4 mA)
Forward flow
Reverse flow

• The current output signal is independent of the direction of flow (absolute amount of themeasured variable). The values for the Lower range value output parameter (→  77)
and Upper range value output parameter (→  78) must have the same sign.
• The value for the Upper range value output parameter (→  78) (e.g. reverse flow)corresponds to the mirrored value for the Upper range value output parameter(→  78) (e.g. forward flow).
"Reverse flow compensation" option
The Reverse flow compensation option is primarily used to compensate for intermittent
reverse flow that can arise with displacement pumps due to wear or high-viscosity
medium. The reverse flow is recorded in a buffer memory and offset against the next
forward flow.
In the event of prolonged and undesired reverse flow, flow values can accumulate in thebuffer memory. Due to the configuration of the current output, these values are notfactored in, however, i.e. there is no compensation for the reverse flow.

*

Visibility depends on order options or device settings

Endress+Hauser

79Description of device parameters

Proline Teqwave M 300 HART

If this option is set, the measuring device does not smoothen the flow signal. The flow
signal is not attenuated.
Examples of how the current output behaves
Example 1
Defined measuring range: lower range value and upper range value with the same sign
I [mA]
20

4
0

1

2

Q
A0028084

2
I
Q
1
2

Measuring range
Current
Flow
Lower range value (Start of measuring range output)
Upper range value (end of measuring range output)

With the following flow response:Qt
A0028091

3
Qt

Flow response
Flow
Time

With Forward flow option
The current output signal is proportional to the process variable assigned. The flow
components outside the scaled measuring range are not taken into account for signal
output:.
I [mA]

t
A0028092

It

Current
Time

With Forward/Reverse flow option
The current output signal is independent of the direction of flow.

80

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

I [mA]

t
A0028093

It

Current
Time

With Reverse flow compensation option
Flow components outside the measuring span are buffered, balanced and output after amaximum delay of 60 s.
I [mA]At

S
S=A

A0028094

It
S
A

Current
Time
Flow components saved
Balancing of saved flow components

Example 2
Defined measuring range: lower range value and upper range value with different signs
I [mA]
20
4
0

Q

0
1

2
A0028095

4
I
Q
1
2

Measuring range
Current
Flow
Lower range value (Start of measuring range output)
Upper range value (end of measuring range output)

With flow a (—) outside, b (- -) inside the measuring range

Endress+Hauser

81Description of device parameters

Proline Teqwave M 300 HART

Q
1
+

t

0
2

-

a=
b=
A0028098

Q
t
1
2

Flow
Time
Lower range value (Start of measuring range output)
Upper range value (end of measuring range output)

With Forward flow option• a (—): The flow components outside the scaled measuring range cannot be taken into
account for signal output.
• b (- -): The current output signal is proportional to the process variable assigned.
I [mA]ta=b=
A0028100It

Current
Time

With Forward/Reverse flow option
This option cannot be selected here since the values for the Lower range value output
parameter (→  77) and Upper range value output parameter (→  78) have different
signs.
With Reverse flow compensation option
Flow components outside the measuring span are buffered, balanced and output after amaximum delay of 60 s.
I [mA]At

S
S=A

A0028101

I
t
S
A

82

Current
Time
Flow components saved
Balancing of saved flow components

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Damping current output

Navigation

 Expert → Output → Curr.output 1 to n → Damp.curr.outp

Prerequisite

A process variable is selected in the Assign current output parameter (→  75) and oneof the following options is selected in the Current span parameter (→  75):
• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4...20.5 mA)• 0...20 mA (0...20.5 mA)

Description

Use this function to enter a time constant for the reaction time of the current output signal
to fluctuations in the measured value caused by process conditions.

User entry

0.0 to 999.9 s

Factory setting

1.0 s

Additional information

User entry
Use this function to enter a time constant (PT1 element 2)) for current output damping:• If a low time constant is entered, the current output reacts quickly to fluctuating
measured variables.
• If a high time constant is entered, the current output reacts more slowly.
Damping is switched off if 0 is entered (factory setting).

Failure behavior current output

Navigation

 Expert → Output → Curr.output 1 to n → Failure behav.

Prerequisite

A process variable is selected in the Assign current output parameter (→  75) and oneof the following options is selected in the Current span parameter (→  75):
• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)• 4...20 mA (4...20.5 mA)• 0...20 mA (0...20.5 mA)

Description

Use this function to select the value of the current output in the event of a device alarm.

Selection• Min.
• Max.
• Last valid value• Actual value• Fixed value

Factory setting

Max.

2)

proportional transmission behavior with first order delay

Endress+Hauser

83Description of device parameters

Additional information

Proline Teqwave M 300 HART

Description
This setting does not affect the failsafe mode of other outputs and totalizers. This isspecified in separate parameters.
"Min." option
The current output adopts the value of the lower level for signal on alarm.
The signal on alarm level is defined via the Current span parameter (→  75).
"Max." option
The current output adopts the value of the upper level for signal on alarm.
The signal on alarm level is defined via the Current span parameter (→  75).
"Last valid value" option
The current output adopts the last measured value that was valid before the device alarm
occurred.
"Actual value" option
The current output adopts the measured value on the basis of the current flow
measurement; the device alarm is ignored.
"Defined value" option
The current output adopts a defined measured value.
The measured value is defined via the Failure current parameter (→  84).

Failure current

Navigation

 Expert → Output → Curr.output 1 to n → Fail. current

Prerequisite

The Defined value option is selected in the Failure mode parameter (→  83).

Description

Use this function to enter a fixed value that the current output adopts in the event of adevice alarm.

User entry

0 to 22.5 mA

Factory setting

22.5 mA

Output current
Navigation

 Expert → Output → Curr.output 1 to n → Output curr.

Description

Displays the current value currently calculated for the current output.

User interface

3.59 to 22.5 mA

84

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Measured current
Navigation

 Expert → Output → Curr.output 1 to n → Measur. curr.

Description

Displays the actual measured value of the output current.

User interface

0 to 30 mA

3.5.2
Navigation"Pulse/frequency/switch output 1 to n" submenu
 Expert → Output → PFS output 1 to n

‣ Pulse/frequency/switch output
1 to nEndress+Hauser

Terminal number→  86Signal mode→  87

Operating mode→  87

Assign pulse output→  88

Pulse scaling→  89Pulse width→  89Measuring mode

→  90

Failure mode

→  91

Pulse output→  91

Assign frequency output→  92

Minimum frequency value→  92

Maximum frequency value→  93

Measuring value at minimum
frequency→  93

Measuring value at maximum
frequency→  93

Measuring mode

→  9485

Description of device parameters

Proline Teqwave M 300 HART

Damping output→  94

Response time→  95

Failure mode

→  95

Failure frequency→  96

Output frequency→  96

Switch output function→  96

Assign diagnostic behavior→  97

Assign limit→  98Switch-on value→  100

Switch-off value→  100

Assign flow direction check→  101

Assign status→  101

Switch-on delay→  101

Switch-off delay→  102

Failure mode→  102

Switch state→  102

Invert output signal→  103

Terminal number
Navigation

 Expert → Output → PFS output 1 to n → Terminal no.

Description

Displays the terminal numbers used by the pulse/frequency/switch output module.

User interface• Not used
• 24-25 (I/O 2)• 22-23 (I/O 3)

Additional information"Not used" option
The pulse/frequency/switch output module does not use any terminal numbers.

86

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Signal modeNavigation

 Expert → Output → PFS output 1 to n → Signal mode

Description

Use this function to select the signal mode for the pulse/frequency/switch output.

Selection• Passive• Active *• Passive NE

Factory setting

Passive

Operating modeNavigation

 Expert → Output → PFS output 1 to n → Operating mode

Prerequisite

If the Pulse option is selected, the Load rate option must be selected in the Assign pulse
output parameter (→  88).

Description

Use this function to select the operating mode of the output as a pulse, frequency or switch
output.

Selection• Pulse• Frequency• Switch

Factory setting

Pulse

Additional information"Pulse" option
Example• Flow rate approx. 100 g/s• Pulse value0.1 g• Pulse width 0.05 ms• Pulse rate 1 000 Impuls/s*

Visibility depends on order options or device settings

Endress+Hauser

87Description of device parameters

Proline Teqwave M 300 HART

U[V]

B<P
B

24
t

0
P

A0026883

5
BP

Quantity-proportional pulse (pulse value) with pulse width to be configured
Pulse width entered
Pauses between the individual pulses"Frequency" option
Example• Flow rate approx. 100 g/s• Max. frequency 10 kHz• Flow rate at max. frequency 1 000 g/s• Output frequency approx. 1 000 Hz
U[V]

fout
T!T"

T!= T"

24
0tA0026886

6

Flow-proportional frequency output

Assign pulse output

Navigation

 Expert → Output → PFS output 1 to n → Assign pulse

Prerequisite

The Load rate option is only available if the volume flow of the medium is read in via theCurrent input 1 to n (→  49) or the fieldbus.

Prerequisite

The Pulse option is selected in Operating mode parameter (→  87).

Description

Use this function to select the process variable for the pulse output.

Selection• Off• Load rate *Factory setting

Off

*

88

Visibility depends on order options or device settings

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Pulse scaling

Navigation

 Expert → Output → PFS output 1 to n → Pulse scaling

Prerequisite

The Pulse option is selected in the Operating mode parameter (→  87) and a process
variable is selected in the Assign pulse output parameter (→  88).

Description

Use this function to enter the value for the measured value that a pulse is equivalent to.

User entry

Positive floating point number

Factory setting

Depends on country and nominal diameter

Additional information

User entry
Weighting of the pulse output with a quantity.
The lower the pulse value, the• better the resolution.
• the higher the frequency of the pulse response.

Pulse widthNavigation

 Expert → Output → PFS output 1 to n → Pulse width

Prerequisite

The Pulse option is selected in the Operating mode parameter (→  87) and a process
variable is selected in the Assign pulse output parameter (→  88).

Description

Use this function to enter the duration of the output pulse.

User entry

0.05 to 2 000 ms

Factory setting

100 msAdditional information

Description• Define how long a pulse is (duration).
• The maximum pulse rate is defined by fmax = 1 / (2 × pulse width).
• The interval between two pulses lasts at least as long as the set pulse width.
• The maximum flow is defined by Qmax = fmax × pulse value.
• If the flow exceeds these limit values, the measuring device displays the 443 Pulse
output 1 to n diagnostic message.

Endress+Hauser

89Description of device parameters

Proline Teqwave M 300 HART

U[V]

U[V]

B<P

B=P
B

B

t
P

t
P
A0026882

B
P

Pulse width entered
Pauses between the individual pulses

Example• Pulse value: 0.1 g• Pulse width: 0.1 ms• fmax: 1 / (2 × 0.1 ms) = 5 kHz• Qmax: 5 kHz × 0.1 g = 0.5 kg/s

Measuring modeNavigation

 Expert → Output → PFS output 1 to n → Measuring mode

Description

Use this function to select the measuring mode for the pulse output.

Selection• Forward flow
• Forward/Reverse flow
• Reverse flow
• Reverse flow compensation

Factory setting

Forward flow

Additional information

Options• Forward flow
Positive flow is output, negative flow is not output.
• Forward/Reverse flow
Positive and negative flow are output (absolute value), but a distinction is not madebetween positive and negative flow.
• Reverse flow
Negative flow is output, positive flow is not output.
• Reverse flow compensation
The flow components outside the measuring range are buffered, balanced and output
after a maximum delay of 60 s.
For a detailed description of the options available, see the Measuring modeparameter (→  78)
ExamplesFor a detailed description of the configuration examples, see the Measuring modeparameter (→  78)

90

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Failure modeNavigation

 Expert → Output → PFS output 1 to n → Failure mode

Prerequisite

The Pulse option is selected in the Operating mode parameter (→  87) and a process
variable is selected in the Assign pulse output parameter (→  88).

Description

Use this function to select the failure mode of the pulse output in the event of a device
alarm.

Selection• Actual value• No pulses

Factory setting

No pulses

Additional information

Description
The dictates of safety render it advisable to ensure that the pulse output shows apredefined behavior in the event of a device alarm.
Options• Actual value
In the event of a device alarm, the pulse output continues on the basis of the current
flow measurement. The fault is ignored.
• No pulses
In the event of a device alarm, the pulse output is "switched off".
NOTICE! A device alarm indicates a serious fault with the measuring device. The
measurement quality may possibly be influenced and may no longer be guaranteed. The
Actual value option is only recommended if it is ensured that all possible alarm conditions
do not influence the measurement quality.

Pulse output
Navigation

 Expert → Output → PFS output 1 to n → Pulse output

Prerequisite

The Pulse option is selected in the Operating mode parameter (→  87) parameter.

Description

Displays the pulse frequency currently output.

User interface

Positive floating-point number

Additional information

Description• The pulse output is an open collector output.
• This is configured at the factory in such a way that the transistor is conductive for theduration of the pulse (NO contact) and is safety-oriented.

Endress+Hauser

91Description of device parameters

Proline Teqwave M 300 HART

1

1

t

0

t

0

NC

NO
A0028726

0
1
NC
NO

Non-conductive
Conductive
NC contact (normally closed)
NO contact (normally open)

The output behavior can be reversed via the Invert output signal parameter (→  103)
i.e. the transistor does not conduct for the duration of the pulse.
In addition, the behavior of the output in the event of a device alarm (Failure modeparameter (→  91)) can be configured.

Assign frequency output

Navigation

 Expert → Output → PFS output 1 to n → Assign freq.

Prerequisite• The Frequency option is selected in Operating mode parameter (→  87).
• The Load rate option is only available if the volume flow of the medium is read in via theCurrent input 1 to n (→  49) or the fieldbus.

Description

Use this function to select the process variable for the frequency output.

Selection• Off• Total solids• Temperature• Electronics temperature• Conductivity• Corrected conductivity• Load rate *Factory setting

Off

Minimum frequency value

Navigation

 Expert → Output → PFS output 1 to n → Min. freq. value

Prerequisite

The Frequency option is selected in the Operating mode parameter (→  87) and aprocess variable is selected in the Assign frequency output parameter (→  92).

Description

Use this function to enter the minimum frequency.

User entry

0.0 to 10 000.0 Hz*

92

Visibility depends on order options or device settings

Endress+Hauser

Proline Teqwave M 300 HART

Factory setting

Description of device parameters

0.0 HzMaximum frequency value

Navigation

 Expert → Output → PFS output 1 to n → Max. freq. value

Prerequisite

The Frequency option is selected in the Operating mode parameter (→  87) and aprocess variable is selected in the Assign frequency output parameter (→  92).

Description

Use this function to enter the end value frequency.

User entry

0.0 to 10 000.0 Hz

Factory setting

10 000.0 Hz

Measuring value at minimum frequency
Navigation

 Expert → Output → PFS output 1 to n → Val. at min.freq

Prerequisite

The Frequency option is selected in the Operating mode parameter (→  87) and aprocess variable is selected in the Assign frequency output parameter (→  92).

Description

Use this function to enter the measured value for the start value frequency.

User entry

Signed floating-point number

Factory setting

Depends on country and nominal diameter

Additional information

DependencyThe entry depends on the process variable selected in the Assign frequency output
parameter (→  92).

Measuring value at maximum frequency
Navigation

 Expert → Output → PFS output 1 to n → Val. at max.freq

Prerequisite

The Frequency option is selected in the Operating mode parameter (→  87) and aprocess variable is selected in the Assign frequency output parameter (→  92).

Description

Use this function to enter the measured value for the end value frequency.

User entry

Signed floating-point number

Factory setting

Depends on country and nominal diameter

Endress+Hauser93Description of device parameters

Additional information

Proline Teqwave M 300 HART

Description
Use this function to enter the maximum measured value at the maximum frequency. The
selected process variable is output as a proportional frequency.
Dependency
The entry depends on the process variable selected in the Assign frequency output
parameter (→  92).

Measuring modeNavigation

 Expert → Output → PFS output 1 to n → Measuring mode

Description

Use this function to select the measuring mode for the frequency output.

Selection• Forward flow
• Forward/Reverse flow
• Reverse flow compensation

Factory setting

Forward flow

Additional information

Options
For a detailed description of the options available, see the Measuring modeparameter (→  78)
ExamplesFor a detailed description of the configuration examples, see the Measuring modeparameter (→  78)

Damping output

Navigation

 Expert → Output → PFS output 1 to n → Damping out.

Description

Use this function to enter a time constant for the reaction time of the output signal tofluctuations in the measured value.

User entry

0 to 999.9 s

Factory setting

0.0 s

94

Endress+Hauser

Proline Teqwave M 300 HART

Additional information

Description of device parameters

User entry
Use this function to enter a time constant (PT1 element 3)) for frequency output damping:• If a low time constant is entered, the current output reacts particularly quickly tofluctuating measured variables.
• On the other hand, the current output reacts more slowly if a high time constant isentered.
Damping is switched off if 0 is entered (factory setting).
The frequency output is subject to separate damping that is independent of all precedingtime constants.

Response time
Navigation

 Expert → Output → PFS output 1 to n → Response time

Prerequisite

The Frequency option is selected in the Operating mode parameter (→  87) and an
option is selected in the Assign frequency output parameter (→  92).

Description

Shows how quickly the output reaches 63% of a measured value change.

User interface

Positive floating-point number

Factory setting

0s

Failure modeNavigation

 Expert → Output → PFS output 1 to n → Failure mode

Prerequisite

The Frequency option is selected in the Operating mode parameter (→  87) and aprocess variable is selected in the Assign frequency output parameter (→  92).

Description

Use this function to select the failure mode of the frequency output in the event of a device
alarm.

Selection• Actual value• Defined value• 0 HzFactory setting

0 Hz

3)

proportional transmission behavior with first order delay

Endress+Hauser

95Description of device parameters

Additional information

Proline Teqwave M 300 HART

Options• Actual value
In the event of a device alarm, the frequency output continues on the basis of the current
flow measurement. The device alarm is ignored.
• Defined value
In the event of a device alarm, the frequency output continues on the basis of apredefined value. The Failure frequency (→  96) replaces the current measured
value, making it possible to bypass the device alarm. The actual measurement isswitched off for the duration of the device alarm.
• 0 HzIn the event of a device alarm, the frequency output is "switched off".
NOTICE! A device alarm indicates a serious fault with the measuring device. The
measurement quality may possibly be influenced and may no longer be guaranteed. The
Actual value option is only recommended if it is ensured that all possible alarm conditions
do not influence the measurement quality.

Failure frequency

Navigation

 Expert → Output → PFS output 1 to n → Failure freq.

Prerequisite

In the Operating mode parameter (→  87), the Frequency option is selected, in the
Assign frequency output parameter (→  92) a process variable is selected, and in the
Failure mode parameter (→  95), the Defined value option is selected.

Description

Enter frequency output value in alarm condition.

User entry

0.0 to 12 500.0 Hz

Factory setting

0.0 HzOutput frequency
Navigation

 Expert → Output → PFS output 1 to n → Output freq.

Prerequisite

In the Operating mode parameter (→  87), the Frequency option is selected.

Description

Displays the actual value of the output frequency which is currently measured.

User interface

0.0 to 12 500.0 Hz

Switch output function



Navigation

 Expert → Output → PFS output 1 to n → Switch out funct

Prerequisite

The Switch option is selected in the Operating mode parameter (→  87).

96

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Description

Use this function to select a function for the switch output.

Selection

• Off
• On
• Diagnostic behavior• Limit• Flow direction check• Status

Factory setting

Off

Additional information• Off
The switch output is permanently switched off (open, non-conductive).
• On
The switch output is permanently switched on (closed, conductive).
• Diagnostic behavior
The switch output is switched on (closed, conductive), if there is a pending diagnostic
event of the assigned behavioral category.
• Limit
The switch output is switched on (closed, conductive), if a limit value specified for theprocess variable is reached.
• Flow direction check
The switch output is switched on (closed, conductive), when the flow direction changes(forward or reverse flow).
• Status
The switch output is switched on (closed/conductive) to display the device status for theselected detection method, e.g. empty pipe detection.

Assign diagnostic behavior

Navigation

 Expert → Output → PFS output 1 to n → Assign diag. beh

Prerequisite• In the Operating mode parameter (→  87), the Switch option is selected.
• In the Switch output function parameter (→  96), the Diagnostic behavior option is
selected.

Description

Use this function to select the diagnostic event category that is displayed for the switch
output.

Selection• Alarm• Alarm or warning• Warning

Factory setting

Alarm

Endress+Hauser

97Description of device parameters

Additional information

Proline Teqwave M 300 HART

Description
If no diagnostic event is pending, the switch output is closed and conductive.
Selection• Alarm
The switch output signals only diagnostic events in the alarm category.
• Alarm or warning
The switch output signals diagnostic events in the alarm and warning category.
• Warning
The switch output signals only diagnostic events in the warning category.

Assign limitNavigation

 Expert → Output → PFS output 1 to n → Assign limit

Prerequisite• The Switch option is selected in Operating mode parameter (→  87).
• The Limit option is selected in Switch output function parameter (→  96).
• The Load rate option is only available if the volume flow of the medium is read in via theCurrent input 1 to n (→  49) or the fieldbus.

Description

Use this function to select a process variable for the limit function.

Selection• Off• Total solids• Temperature• Electronics temperature• Conductivity• Corrected conductivity• Load rate *• Totalizer 1 *Factory setting

Temperature

Additional information

Description
Behavior of status output when Switch-on value > Switch-off value:• Process variable > Switch-on value: transistor is conductive• Process variable < Switch-off value: transistor is non-conductive*

98

Visibility depends on order options or device settings

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

A
1
2
t
B
3
4

t
A0026891

1
2
3
4
A
B

Switch-on value
Switch-off value
Conductive
Non-conductive
Process variable
Status output

Behavior of status output when Switch-on value < Switch-off value:• Process variable < Switch-on value: transistor is conductive• Process variable > Switch-off value: transistor is non-conductive
A
1
2
t
B
3
4

t
A0026892

1
2
3
4
A
B

Switch-off value
Switch-on value
Conductive
Non-conductive
Process variable
Status output

Behavior of status output when Switch-on value = Switch-off value:• Process variable > Switch-on value: transistor is conductive• Process variable < Switch-off value: transistor is non-conductive
A
1
t
B
2
3

t
A0026893

1
2
3
A
B

Endress+Hauser

Switch-on value = Switch-off value
Conductive
Non-conductive
Process variable
Status output

99

Description of device parameters

Proline Teqwave M 300 HART

Switch-on value

Navigation

 Expert → Output → PFS output 1 to n → Switch-on value

Prerequisite• The Switch option is selected in the Operating mode parameter (→  87).
• The Limit option is selected in the Switch output function parameter (→  96).

Description

Use this function to enter the measured value for the switch-on point.

User entry

Floating point number with sign

Factory setting

0 °C

Additional information

Description
Use this function to enter the limit value for the switch-on value (process variable >switch-on value = closed, conductive).
When using a hysteresis: Switch-on value > Switch-off value.
Dependency
The unit depends on the process variable selected in the Assign limit parameter(→  98).

Switch-off value

Navigation

 Expert → Output → PFS output 1 to n → Switch-off value

Prerequisite• The Switch option is selected in the Operating mode parameter (→  87).
• The Limit option is selected in the Switch output function parameter (→  96).

Description

Use this function to enter the measured value for the switch-off point.

User entry

Floating point number with sign

Factory setting

0 °C

Additional information

Description
Use this function to enter the limit value for the switch-off value (process variable <switch-off value = open, non-conductive).
When using a hysteresis: Switch-on value > Switch-off value.
Dependency
The unit depends on the process variable selected in the Assign limit parameter(→  98).

100

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Assign flow direction checkNavigation

 Expert → Output → PFS output 1 to n → Assign dir.check

Prerequisite• The Switch option is selected in the Operating mode parameter (→  87).
• The Flow direction check option is selected in the Switch output function parameter(→  96).

Description

Use this function to select a process variable for monitoring the flow direction.

Selection
Factory setting

Off

Assign status



Navigation

 Expert → Output → PFS output 1 to n → Assign status

Prerequisite• The Switch option is selected in Operating mode parameter (→  87).
• The Status option is selected in Switch output function parameter (→  96).

Description

Select the device function whose status you want to display.

Selection• Off• Partially filled pipe detection

Factory setting

Partially filled pipe detection

Additional information

Options
When the switch-on point for the selected device function is reached, the output isswitched on (closed, conductive). Otherwise, the output is non-conductive.

Switch-on delay

Navigation

 Expert → Output → PFS output 1 to n → Switch-on delay

Prerequisite• The Switch option is selected in the Operating mode parameter (→  87).
• The Limit option is selected in the Switch output function parameter (→  96).

Description

Use this function to enter a delay time for switching on the switch output.

User entry

0.0 to 100.0 s

Factory setting

0.0 s

Endress+Hauser

101Description of device parameters

Proline Teqwave M 300 HART

Switch-off delay

Navigation

 Expert → Output → PFS output 1 to n → Switch-off delay

Prerequisite• The Switch option is selected in the Operating mode parameter (→  87).
• The Limit option is selected in the Switch output function parameter (→  96).

Description

Use this function to enter a delay time for switching off the switch output.

User entry

0.0 to 100.0 s

Factory setting

0.0 s

Failure modeNavigation

 Expert → Output → PFS output 1 to n → Failure mode

Description

Use this function to select a failsafe mode for the switch output in the event of a device
alarm.

Selection• Actual status• Open• Closed

Factory setting

Open

Additional information

Options• Actual status
In the event of a device alarm, faults are ignored and the current behavior of the input
value is output by the switch output. The Actual status option behaves in the same way
as the current input value.
• OpenIn the event of a device alarm, the switch output's transistor is set to non-conductive.
• ClosedIn the event of a device alarm, the switch output's transistor is set to conductive.

Switch state
Navigation

 Expert → Output → PFS output 1 to n → Switch state

Prerequisite

The Switch option is selected in the Operating mode parameter (→  87).

Description

Displays the current switch status of the status output.

User interface• Open• Closed

102

Endress+Hauser

Proline Teqwave M 300 HART

Additional information

Description of device parameters

User interface• OpenThe switch output is not conductive.
• Closed
The switch output is conductive.

Invert output signal

Navigation

 Expert → Output → PFS output 1 to n → Invert outp.sig.

Description

Use this function to select whether to invert the output signal.

Selection

• No
• Yes

Factory setting

No

Additional information

Selection
No option (passive - negative)

1
0
A0026693

Yes option (passive - positive)

1
0
A0026692

3.5.3
Navigation"Relay output 1 to n" submenu
 Expert → Output → Relay output 1 to n

‣ Relay output 1 to n

Endress+Hauser

Terminal number→  104

Relay output function→  104

Assign flow direction check→  105103Description of device parameters

Proline Teqwave M 300 HART

Assign limit→  105

Assign diagnostic behavior→  106

Assign status→  106

Switch-off value→  107

Switch-off delay→  107

Switch-on value→  107

Switch-on delay→  108

Failure mode→  108

Switch state→  109

Powerless relay status→  109

Terminal number
Navigation

 Expert → Output → Relay output 1 to n → Terminal no.

Description

Displays the terminal numbers used by the relay output module.

User interface• Not used
• 24-25 (I/O 2)• 22-23 (I/O 3)

Additional information"Not used" option
The relay output module does not use any terminal numbers.

Relay output function



Navigation

 Expert → Output → Relay output 1 to n → Relay outp.func.

Description

Use this function to select an output function for the relay output.

Selection• Closed• Open• Diagnostic behavior• Limit• Flow direction check• Status

104

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Factory setting

ClosedAdditional information

Selection• Closed
The relay output is permanently switched on (closed, conductive).
• OpenThe relay output is permanently switched off (open, non-conductive).
• Diagnostic behavior
Indicates if the diagnostic event is present or not. Is used to output diagnostic
information and to react to it appropriately at the system level.
• Limit
Indicates if a specified limit value has been reached for the process variable. Is used to
output diagnostic information relating to the process and to react to it appropriately at
the system level.
• Flow direction check
Indicates the flow direction (forward or reverse flow).
• Digital Output
Indicates the device status depending on whether empty pipe detection or low flow cut
off is selected.

Assign flow direction checkNavigation

 Expert → Output → Relay output 1 to n → Assign dir.check

Prerequisite

The Flow direction check option is selected in the Relay output function parameter(→  104).

Description

Use this function to select a process variable for monitoring the flow direction.

Selection
Factory setting

Off

Assign limitNavigation

 Expert → Output → Relay output 1 to n → Assign limit

Prerequisite• The Limit option is selected in Relay output function parameter (→  104).
• The Load rate option is only available if the volume flow of the medium is read in via theCurrent input 1 to n (→  49) or the fieldbus.

Description

Use this function to select a process variable for the limit value function.

Selection• Off• Total solids• Temperature• Electronics temperature• Conductivity

Endress+Hauser

105Description of device parameters

Proline Teqwave M 300 HART

• Corrected conductivity• Load rate *• Totalizer 1 *Factory setting

Temperature

Assign diagnostic behavior

Navigation

 Expert → Output → Relay output 1 to n → Assign diag. beh

Prerequisite

In the Relay output function parameter (→  104), the Diagnostic behavior option is
selected.

Description

Use this function to select the category of the diagnostic events that are displayed for therelay output.

Selection• Alarm• Alarm or warning• Warning

Factory setting

Alarm

Additional information

Description
If no diagnostic event is pending, the relay output is closed and conductive.
Selection• Alarm
The relay output signals only diagnostic events in the alarm category.
• Alarm or warning
The relay output signals diagnostic events in the alarm and warning category.
• Warning
The relay output signals only diagnostic events in the warning category.

Assign status



Navigation

 Expert → Output → Relay output 1 to n → Assign status

Prerequisite

In the Relay output function parameter (→  104), the Digital Output option is
selected.

Description

Use this function to select the device status for the relay output.

Selection• Off• Partially filled pipe detection

Factory setting

Off

*

106

Visibility depends on order options or device settings

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Switch-off value

Navigation

 Expert → Output → Relay output 1 to n → Switch-off value

Prerequisite

The Limit option is selected in the Relay output function parameter (→  104).

Description

Use this function to enter the measured value for the switch-off point.

User entry

Floating point number with sign

Factory setting

0 °C

Additional information

Description
Use this function to enter the limit value for the switch-off value (process variable <switch-off value = open, non-conductive).
When using a hysteresis: Switch-on value > Switch-off value.
Dependency
The unit is dependent on the process variable selected in the Assign limit parameter(→  105).

Switch-off delay

Navigation

 Expert → Output → Relay output 1 to n → Switch-off delay

Prerequisite

In the Relay output function parameter (→  104), the Limit option is selected.

Description

Use this function to enter a delay time for switching off the switch output.

User entry

0.0 to 100.0 s

Factory setting

0.0 s

Switch-on value

Navigation

 Expert → Output → Relay output 1 to n → Switch-on value

Prerequisite

The Limit option is selected in the Relay output function parameter (→  104).

Description

Use this function to enter the measured value for the switch-on point.

User entry

Floating point number with sign

Factory setting

0 °C

Endress+Hauser

107Description of device parameters

Additional information

Proline Teqwave M 300 HART

Description
Use this function to enter the limit value for the switch-on value (process variable >switch-on value = closed, conductive).
When using a hysteresis: Switch-on value > Switch-off value.
Dependency
The unit is dependent on the process variable selected in the Assign limit parameter(→  105).

Switch-on delay

Navigation

 Expert → Output → Relay output 1 to n → Switch-on delay

Prerequisite

In the Relay output function parameter (→  104), the Limit option is selected.

Description

Use this function to enter a delay time for switching on the switch output.

User entry

0.0 to 100.0 s

Factory setting

0.0 s

Failure modeNavigation

 Expert → Output → Relay output 1 to n → Failure mode

Description

Use this function to select the failure mode of the relay output in the event of a device
alarm.

Selection• Actual status• Open• Closed

Factory setting

Open

Additional information

Selection• Actual status
In the event of a device alarm, faults are ignored and the current behavior of the input
value is output by the relay output. The Actual status option behaves in the same way as
the current input value.
• OpenIn the event of a device alarm, the relay output's transistor is set to non-conductive.
• ClosedIn the event of a device alarm, the relay output's transistor is set to conductive.

108

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Switch state
Navigation

 Expert → Output → Relay output 1 to n → Switch state

Description

Displays the current status of the relay output.

User interface• Open• Closed

Additional information

User interface• OpenThe relay output is not conductive.
• Closed
The relay output is conductive.

Powerless relay status



Navigation

 Expert → Output → Relay output 1 to n → Powerless relay

Description

Use this function to select the quiescent state for the relay output.

Selection• Open• Closed

Factory setting

Open

Additional information

Selection• OpenThe relay output is not conductive.
• Closed
The relay output is conductive.

3.6"Communication" submenu

Navigation

 Expert → Communication‣ Communication

Endress+Hauser‣ HART input→  110‣ HART output→  115‣ Diagnostic configuration→  131‣ Web server→  135

109

Description of device parameters

Proline Teqwave M 300 HART

3.6.1

"HART input" submenu

Navigation

 Expert → Communication → HART input‣ HART input‣ Configuration→  110‣ Input→  114"Configuration" submenu
Navigation

 Expert → Communication → HART input → Configuration‣ Configuration
Capture mode→  110Device ID→  111

Device type→  111

Manufacturer ID→  112

Burst command→  112

Slot number→  113

Timeout→  113

Failure mode→  113

Failure value→  114

Capture modeNavigation

 Expert → Communication → HART input → Configuration → Capture mode

Description

Use this function to select the capture mode via burst or master communication.

Selection• Off• Burst network• Master network

Factory setting

Off

110

Endress+Hauser

Proline Teqwave M 300 HART

Additional information

Description of device parameters"Burst network" option
The device records data transmitted via burst in the network.
"Master network" option
In this case, the device must be located in a HART network in which a HART master(control) queries the measured values of the up to 64 network participants. The device
reacts only to the responses of a specific device in the network. Device ID,device type,manufacturer ID and the HART commands used by the master must be defined.

Device ID

Navigation

 Expert → Communication → HART input → Configuration → Device ID

Prerequisite

The Master network option is selected in the Capture mode parameter (→  110).

Description

Use this function to enter the device ID of the HART slave device whose data are to berecorded.

User entry

6-digit value:• Via local operation: enter as hexadecimal or decimal number• Via operating tool: enter as decimal number

Factory setting0

Additional information

In addition to the device ID and manufacturer ID, the device type is part of the unique
ID. Each HART device is uniquely identified by the unique device ID.

Device type

Navigation

 Expert → Communication → HART input → Configuration → Device type

Prerequisite

In the Capture mode parameter (→  110), the Master network option is selected.

Description

Use this function to enter the device type of the HART slave device whose data are to berecorded.

User entry

2-digit hexadecimal number

Factory setting

0x00

Additional information

Endress+Hauser

In addition to the device ID and manufacturer ID, the device type is part of the unique
ID. Each HART device is uniquely identified by the unique device ID.

111

Description of device parameters

Proline Teqwave M 300 HART

Manufacturer ID

Navigation

 Expert → Communication → HART input → Configuration → Manufacturer ID

Prerequisite

The Master network option is selected in the Capture mode parameter (→  110).

Description

Use this function to enter the manufacturer ID of the HART slave device whose data are to
be recorded.

User entry

2-digit value:• Via local operation: enter as hexadecimal or decimal number• Via operating tool: enter as decimal number

Factory setting0

Additional information

In addition to the device ID and manufacturer ID, the device type is part of the unique
ID. Each HART device is uniquely identified by the unique device ID.

Burst commandNavigation

 Expert → Communication → HART input → Configuration → Burst command

Prerequisite

The Burst network option or the Master network option are selected in the Capturemode parameter (→  110).

Description

Use this function to select the burst command to be recorded.

Selection• Command 1• Command 3• Command 9• Command 33

Factory setting

Command 1

Additional information

Selection• Command 1
Use this function to capture the primary variable.
• Command 3
Use this function to capture the dynamic HART variables and the current.
• Command 9
Use this function to capture the dynamic HART variables including the associated status.
• Command 33
Use this function to capture the dynamic HART variables including the associated unit.

112

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Slot number



Navigation

 Expert → Communication → HART input → Configuration → Slot number

Prerequisite

The Burst network option or the Master network option is selected in the Capture modeparameter (→  110).

Description

Use this function to enter the position of the process variable to be recorded in the burst
command.

User entry

1 to 8

Factory setting

1

Additional information

User entry
Slot

Command
1

3

9

33

1

PV

PV

HART variable (slot 1)

HART variable (slot 1)

2

–

SV

HART variable (slot 2)

HART variable (slot 2)

3

–

TV

HART variable (slot 3)

HART variable (slot 3)

4

–

QV

HART variable (slot 4)

HART variable (slot 4)

Timeout

Navigation

 Expert → Communication → HART input → Configuration → Timeout

Prerequisite

The Burst network option or the Master network option is selected in the Capture modeparameter (→  110).

Description

Use this function to enter the maximum permitted interval between two HART frames.

User entry

1 to 120 s

Factory setting

5sAdditional information

Description
If the interval is exceeded, the measuring device displays the F882 Input signal
diagnostic message.

Failure modeNavigation

 Expert → Communication → HART input → Configuration → Failure mode

Prerequisite

In the Capture mode parameter (→  110), the Burst network option or Master
network option is selected.

Endress+Hauser

113

Description of device parameters

Proline Teqwave M 300 HART

Description

Use this function to select the device behavior if no data are recorded within the maximum
permitted interval.

Selection• Alarm• Last valid value• Defined value

Factory setting

Alarm

Additional information

Options• Alarm
An error message is set.
• Last valid value
The last valid measured value is used.
• Defined value
A user-defined measured value is used: (Failure value parameter (→  114)).

Failure value

Navigation

 Expert → Communication → HART input → Configuration → Failure value

Prerequisite

The following conditions are met:
• In the Capture mode parameter (→  110), the Burst network option or Master
network option is selected.
• In the Failure mode parameter (→  113), the Defined value option is selected.

Description

Use this function to enter the measured value to be used if no data are recorded within the
maximum permitted interval.

User entry

Signed floating-point number

Factory setting0"Input" submenu
Navigation

 Expert → Communication → HART input → Input‣ Input

114

Value

→  115

Status→  115

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Value
Navigation

 Expert → Communication → HART input → Input → Value

Description

Displays the value of the device variable recorded by the HART input.

User interface

Signed floating-point number

StatusNavigation

 Expert → Communication → HART input → Input → Status

Description

Displays the value of the device variable recorded by the HART input in accordance with
the HART specification.

User interface• Manual/Fixed• Good• Poor accuracy
• Bad

3.6.2

"HART output" submenu

Navigation

 Expert → Communication → HART output‣ HART output‣ Configuration→  115‣ Burst configuration→  117‣ Information→  124‣ Output→  127"Configuration" submenu
Navigation

 Expert → Communication → HART output → Configuration‣ Configuration

Endress+Hauser

HART short tag→  116

Device tag→  116115

Description of device parameters

Proline Teqwave M 300 HART

HART address→  116

No. of preambles→  117

Fieldbus writing access→  117

HART short tag

Navigation

 Expert → Communication → HART output → Configuration → HART short tag

Description

Use this function to enter a brief description for the measuring point. This can be edited
and displayed via HART protocol or using the local display.

User entry

Max. 8 characters: A to Z, 0 to 9 and certain special characters (e.g. punctuation marks, @,%).

Factory setting

TEQWAVEM

Device tagNavigation

 Expert → Communication → HART output → Configuration → Device tag

Description

Use this function to enter the name for the measuring point.

User entry

Max. 32 characters, such as letters, numbers or special characters (e.g. @, %, /).

Factory setting

Teqwave M

HART addressNavigation

 Expert → Communication → HART output → Configuration → HART address

Description

Use this function to enter the address via which the data exchange takes place via HART
protocol.

User entry

0 to 63

Factory setting0

Additional information

Description
For addressing in a HART Multidrop network, the Fixed current option must be set in the
Current span parameter (→  75) (current output 1).

116

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

No. of preambles



Navigation

 Expert → Communication → HART output → Configuration → No. of preambles

Description

Use this function to enter the number of preambles in the HART protocol.

User entry

2 to 20

Factory setting

5

Additional information

User entry
As every modem component can "swallow" a byte, 2-byte preambles at least must be
defined.

Fieldbus writing access

Navigation

 Expert → Communication → HART output → Configuration → Fieldb.writ.acc.

Description

Use this function to restrict access to the measuring device via fieldbus (HART interface).

Selection• Read + write• Read only

Factory setting

Read + write

Additional information

Description
If read and/or write protection is enabled, the parameter can only be controlled and reset
via local operation. Access is no longer possible via operating tools.
Selection• Read + write
The parameters are readable and writable.
• Read only
The parameters are only readable.

"Burst configuration 1 to n" submenu
Navigation

 Expert → Communication → HART output → Burst config.
→ Burst config. 1 to n

‣ Burst configuration 1 to n

Endress+Hauser

Burst mode 1 to n

→  118

Burst command 1 to n

→  119117

Description of device parameters

Proline Teqwave M 300 HART

Burst variable 0→  119

Burst variable 1→  120

Burst variable 2→  120

Burst variable 3→  121

Burst variable 4→  121

Burst variable 5→  121

Burst variable 6→  121

Burst variable 7→  122

Burst trigger mode→  122

Burst trigger level→  123

Min. update period→  123

Max. update period→  123

Burst mode 1 to n

Navigation

 Expert → Communication → HART output → Burst config. → Burst config. 1 to n
→ Burst mode 1 to n

Description

Use this function to select whether to activate the HART burst mode for burst message X.

Selection

• Off
• On

Factory setting

Off

Additional information

Options• Off
The measuring device transmits data only when requested by the HART master.
• On
The measuring device transmits data regularly without being requested.

118

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Burst command 1 to n

Navigation

 Expert → Communication → HART output → Burst config. → Burst config. 1 to n
→ Burst command 1 to n

Description

Use this function to select the HART command that is sent to the HART master.

Selection• Command 1• Command 2• Command 3• Command 9• Command 33• Command 48

Factory setting

Command 2

Additional information

Selection• Command 1
Read out the primary variable.
• Command 2
Read out the current and the main measured value as a percentage.
• Command 3
Read out the dynamic HART variables and the current.
• Command 9
Read out the dynamic HART variables including the related status.
• Command 33
Read out the dynamic HART variables including the related unit.
• Command 48
Read out the complete device diagnostics.
"Command 33" option
The HART device variables are defined via Command 107.
Commands• Information about the defined details of the command: HART specifications• The measured variables (HART device variables) are assigned to the dynamic
variables in the Output submenu (→  73).

Burst variable 0Navigation

 Expert → Communication → HART output → Burst config. → Burst config. 1 to n
→ Burst variable 0

Prerequisite

The Load rate option is only available if the volume flow of the medium is read in via theCurrent input 1 to n (→  49) or the fieldbus.

Description

For HART command 9 and 33: select the HART device variable or the process variable.

Selection• Conductivity• Corrected conductivity• Temperature• Electronics temperature

Endress+Hauser

119Description of device parameters

Proline Teqwave M 300 HART

• Total solids• Load rate *• Totalizer 1• Percent of range• Measured current• Current input 1 *• Current input 2 *• Current input 3 *• Primary variable (PV)• Secondary variable (SV)• Tertiary variable (TV)• Quaternary variable (QV)
• HART input• Not used
Factory setting

Total solids

Additional information

Options
If a burst message is not configured, the Not used option is set.

Burst variable 1Navigation

 Expert → Communication → HART output → Burst config. → Burst config. 1 to n
→ Burst variable 1

Description

For HART command 9 and 33: select the HART device variable or the process variable.

Selection

See the Burst variable 0 parameter (→  119).

Factory setting

Not used

Burst variable 2

Navigation

 Expert → Communication → HART output → Burst config. → Burst config. 1 to n
→ Burst variable 2

Description

For HART command 9 and 33: select the HART device variable or the process variable.

Selection

See the Burst variable 0 parameter (→  119).

Factory setting

Not used*120

Visibility depends on order options or device settings

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Burst variable 3

Navigation

 Expert → Communication → HART output → Burst config. → Burst config. 1 to n
→ Burst variable 3

Description

For HART command 9 and 33: select the HART device variable or the process variable.

Selection

See the Burst variable 0 parameter (→  119).

Factory setting

Not used

Burst variable 4Navigation

 Expert → Communication → HART output → Burst config. → Burst config. 1 to n
→ Burst variable 4

Description

For HART command 9: select the HART device variable or the process variable.

Selection

See the Burst variable 0 parameter (→  119).

Factory setting

Not used

Burst variable 5Navigation

 Expert → Communication → HART output → Burst config. → Burst config. 1 to n
→ Burst variable 5

Description

For HART command 9: select the HART device variable or the process variable.

Selection

See the Burst variable 0 parameter (→  119).

Factory setting

Not used

Burst variable 6

Navigation

 Expert → Communication → HART output → Burst config. → Burst config. 1 to n
→ Burst variable 6

Description

For HART command 9: select the HART device variable or the process variable.

Selection

See the Burst variable 0 parameter (→  119).

Factory setting

Not used

Endress+Hauser

121

Description of device parameters

Proline Teqwave M 300 HART

Burst variable 7

Navigation

 Expert → Communication → HART output → Burst config. → Burst config. 1 to n
→ Burst variable 7

Description

For HART command 9: select the HART device variable or the process variable.

Selection

See the Burst variable 0 parameter (→  119).

Factory setting

Not used

Burst trigger modeNavigation

 Expert → Communication → HART output → Burst config. → Burst config. 1 to n
→ Trigger mode

Description

Use this function to select the event that triggers burst message X.

Selection• Continuous• Window *• Rising *• Falling *• On change

Factory setting

Continuous

Additional information

Selection• Continuous
The message is sent continuously, at least at intervals corresponding to the time frame
specified in the Burst min period parameter (→  123).
• Window
The message is sent if the specified measured value has changed by the value in the
Burst trigger level parameter (→  123).
• Rising
The message is sent if the specified measured value exceeds the value in the Burst
trigger level parameter (→  123).
• Falling
The message is sent if the specified measured value drops below the value in the Burst
trigger level parameter (→  123).
• On change
The message is sent if a measured value changes in the burst message.

*

122

Visibility depends on order options or device settings

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Burst trigger level

Navigation

 Expert → Communication → HART output → Burst config. → Burst config. 1 to n
→ Trigger level

Description

For entering the burst trigger value.

User entry

Signed floating-point number

Additional information

Description
Together with the option selected in the Burst trigger mode parameter (→  122) the
burst trigger value determines the time of burst message X.

Min. update period

Navigation

 Expert → Communication → HART output → Burst config. → Burst config. 1 to n
→ Min. upd. per.

Description

Use this function to enter the minimum time span between two burst commands of burst
message X.

User entry

Positive integer

Factory setting

1 000 ms

Max. update period

Navigation

 Expert → Communication → HART output → Burst config. → Burst config. 1 to n
→ Max. upd. per.

Description

Use this function to enter the maximum time span between two burst commands of burst
message X.

User entry

Positive integer

Factory setting

2 000 ms

Endress+Hauser

123Description of device parameters

Proline Teqwave M 300 HART

"Information" submenu
Navigation

 Expert → Communication → HART output → Information‣ Information
Device revision→  124

Device ID→  125

Device type→  125

Manufacturer ID→  125

HART revision→  125

HART descriptor→  126

HART message→  126

Hardware revision→  126

Software revision→  126

HART date code→  127

Device revision
Navigation

 Expert → Communication → HART output → Information → Device revision

Description

Displays the device revision with which the device is registered with the HART
Communication Foundation.

User interface

2-digit hexadecimal number

Factory setting

1

Additional information

Description
The device revision is needed to assign the appropriate device description file (DD) tothe device.

124

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Device ID
Navigation

 Expert → Communication → HART output → Information → Device ID

Description

Use this function to view the device ID for identifying the measuring device in a HART
network.

User interface

6-digit hexadecimal number

Additional information

Description
In addition to the device type and manufacturer ID, the device ID is part of the unique
ID. Each HART device is uniquely identified by the unique device ID.

Device type
Navigation

 Expert → Communication → HART output → Information → Device type

Description

Displays the device type used to register the measuring device with the HART
Communication Foundation

User interface

Hexadecimal number

Factory setting

11B3

Additional information

Description
The device type is specified by the manufacturer. It is needed to assign the appropriate
device description file (DD) to the device.

Manufacturer ID
Navigation

 Expert → Communication → HART output → Information → Manufacturer ID

Description

Use this function to view the manufacturer ID with which the measuring device isregistered with the HART Communication Foundation.

User interface

2-digit hexadecimal number

Factory setting

0x11 (for Endress+Hauser)

HART revision
Navigation

 Expert → Communication → HART output → Information → HART revision

Description

Use this function to display the HART protocol revision of the measuring device.

Endress+Hauser

125Description of device parameters

User interface

5 to 7

Factory setting

7

Proline Teqwave M 300 HART

HART descriptor

Navigation

 Expert → Communication → HART output → Information → HART descriptor

Description

Use this function to enter a description for the measuring point. This can be edited anddisplayed via HART protocol or using the local display.

User entry

Max. 16 characters, such as letters, numbers or special characters (e.g. @, %, /)Factory setting

TEQWAVE M300 500

HART message

Navigation

 Expert → Communication → HART output → Information → HART message

Description

Use this function to enter a HART message which is sent via the HART protocol whenrequested by the master.

User entry

Max. 32 characters, such as letters, numbers or special characters (e.g. @, %, /)Factory setting

TEQWAVE M300 500

Hardware revision
Navigation

 Expert → Communication → HART output → Information → Hardware rev.

Description

Displays the hardware revision of the measuring device.

User interface

0 to 255

Factory setting

1

Software revision
Navigation

 Expert → Communication → HART output → Information → Software rev.

Description

Displays the software revision of the measuring device.

User interface

0 to 255

126

Endress+Hauser

Proline Teqwave M 300 HART

Factory setting

Description of device parameters

1

HART date code

Navigation

 Expert → Communication → HART output → Information → HART date code

Description

Use this function to enter the date information for individual use.

User entry

Date entry format: yyyy-mm-dd

Factory setting

2009-07-20

Additional information

ExampleDevice installation date

"Output" submenu
Navigation

 Expert → Communication → HART output → Output‣ OutputEndress+Hauser

Assign PV→  128

Primary variable (PV)→  128

Assign SV→  128

Secondary variable (SV)→  129

Assign TV→  129

Tertiary variable (TV)→  130

Assign QV→  130

Quaternary variable (QV)→  131127

Description of device parameters

Proline Teqwave M 300 HART

Assign PV

Navigation

 Expert → Communication → HART output → Output → Assign PV

Prerequisite

The Load rate option is only available if the volume flow of the medium is read in via theCurrent input 1 to n (→  49) or the fieldbus.

Description

Use this function to select a measured variable (HART device variable) for the primary
dynamic variable (PV).

Selection• Off• Total solids• Temperature• Electronics temperature• Conductivity• Corrected conductivity• Load rate *Factory setting

Total solids

Primary variable (PV)Navigation

 Expert → Communication → HART output → Output → Primary var (PV)

Description

Displays the current measured value of the primary dynamic variable (PV).

User interface

Floating point number with sign

Additional information

Display
The measured value displayed depends on the process variable selected in the Assign PV
parameter (→  128).
Dependency
The unit of the displayed measured value is taken from the System units submenu(→  54).

Assign SVNavigation

 Expert → Communication → HART output → Output → Assign SV

Prerequisite

The Load rate option is only available if the volume flow of the medium is read in via theCurrent input 1 to n (→  49) or the fieldbus.

Description

Use this function to select a measured variable (HART device variable) for the secondary
dynamic variable (SV).

*128

Visibility depends on order options or device settings

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Selection• Conductivity• Corrected conductivity• Temperature• Electronics temperature• Total solids• Load rate *• Volume flow *• Totalizer 1 *• Current input 1 *• Current input 2 *• Current input 3 *• HART input

Factory setting

Temperature

Secondary variable (SV)Navigation

 Expert → Communication → HART output → Output → Second.var(SV)Description

Displays the current measured value of the secondary dynamic variable (SV).

User interface

Floating point number with sign

Additional information

Display
The measured value displayed depends on the process variable selected in the Assign SV
parameter (→  128).
Dependency
The unit of the displayed measured value is taken from the System units submenu(→  54).

Assign TV

Navigation

 Expert → Communication → HART output → Output → Assign TV

Prerequisite

The Load rate option is only available if the volume flow of the medium is read in via theCurrent input 1 to n (→  49) or the fieldbus.

Description

Use this function to select a measured variable (HART device variable) for the tertiary(third) dynamic variable (TV).

Selection• Conductivity• Corrected conductivity• Temperature• Electronics temperature• Total solids• Load rate **Visibility depends on order options or device settings

Endress+Hauser

129

Description of device parameters

Proline Teqwave M 300 HART

• Volume flow *• Totalizer 1 *• Current input 1 *• Current input 2 *• Current input 3 *• HART input
Factory setting

Electronics temperature

Tertiary variable (TV)
Navigation

 Expert → Communication → HART output → Output → Tertiary var(TV)Description

Displays the current measured value of the tertiary dynamic variable (TV).

User interface

Floating point number with sign

Additional information

Display
The measured value displayed depends on the process variable selected in the Assign TV
parameter (→  129).
Dependency
The unit of the displayed measured value is taken from the System units submenu(→  54).

Assign QVNavigation

 Expert → Communication → HART output → Output → Assign QVPrerequisite

The Load rate option is only available if the volume flow of the medium is read in via theCurrent input 1 to n (→  49) or the fieldbus.

Description

Use this function to select a measured variable (HART device variable) for the quaternary(fourth) dynamic variable (QV).

Selection• Conductivity• Corrected conductivity• Temperature• Electronics temperature• Total solids• Load rate *• Volume flow *• Totalizer 1 *• Current input 1 *• Current input 2 *• Current input 3 *• HART input*

130

Visibility depends on order options or device settings

Endress+Hauser

Proline Teqwave M 300 HART

Factory setting

Description of device parameters

Conductivity

Quaternary variable (QV)
Navigation

 Expert → Communication → HART output → Output → Quaterna.var(QV)

Description

Displays the current measured value of the quaternary dynamic variable (QV).

User interface

Floating point number with sign

Additional information

Display
The measured value displayed depends on the process variable selected in the Assign QVparameter (→  130).
Dependency
The unit of the displayed measured value is taken from the System units submenu(→  54).

3.6.3"Diagnostic configuration" submenu

For a list of all the diagnostic events, see the Operating Instructions for the device→ 7
Assign a category to the particular diagnostic event:Category

Meaning

Failure (F)A device error has occurred. The measured value is no longer valid.

Function check (C)The device is in the service mode (e.g. during a simulation).

Out of specification (S)The device is being operated:• Outside its technical specification limits (e.g. outside the process temperature
range)• Outside of the configuration carried out by the user (e.g. maximum flow inparameter 20 mA value)

Maintenance required(M)

Maintenance is required. The measured value remains valid.

No effect (N)Has no effect on the condensed status 1).

1)

Condensed status according to NAMUR recommendation NE 107

Navigation

Expert → Communication → Diag. config.

‣ Diagnostic configuration

Endress+Hauser

Event category 441→  132

Event category 442→  132131

Description of device parameters

Proline Teqwave M 300 HART

Event category 443→  133

Event category 444→  133

Event category 832→  134

Event category 833→  134

Event category 834→  134

Event category 835→  135

Event category 441 (Current output 1 to n)

NavigationDescription

Use this function to select a category for the 441 Current output 1 to n diagnostic
message.

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Factory setting

Out of specification (S)Additional information

Expert → Communication → Diag. config. → Event category 441

For a detailed description of the event categories available for selection: →  131

Event category 442 (Frequency output 1 to n)

Navigation

Prerequisite

The pulse/frequency/switch output is available.

Description

Use this function to select a category for the 442 Frequency output 1 to n diagnostic
message.

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Factory setting

Out of specification (S)132

Expert → Communication → Diag. config. → Event category 442

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Additional information

For a detailed description of the event categories available for selection: →  131

Event category 443 (Pulse output 1 to n)

Navigation

Expert → Communication → Diag. config. → Event category 443

Prerequisite

The pulse/frequency/switch output is available.

Description

Use this function to select a category for the 443 Pulse output 1 to n diagnostic message.

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Factory setting

Out of specification (S)For a detailed description of the event categories available for selection: →  131

Additional information

Event category 444 (Current input 1 to n)

Navigation

Expert → Communication → Diag. config. → Event category 444

Prerequisite

The current input is available.

Description

Use this function to select a category for the 444 Current input 1 to n diagnostic message.

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Factory setting

Out of specification (S)Additional information

Selection
For a detailed description of the event categories available for selection: →  131

Endress+Hauser

133

Description of device parameters

Proline Teqwave M 300 HART

Event category 832 (Electronics temperature too high)

Navigation

Expert → Communication → Diag. config. → Event category 832

Description

Use this function to select a category for the 832 Electronics temperature too highdiagnostic message.

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Factory setting

Out of specification (S)Additional information

Selection
For a detailed description of the event categories available for selection: →  131

Event category 833 (Electronics temperature too low)

Navigation

Expert → Communication → Diag. config. → Event category 833

Description

Use this option to select a category for the 833 Electronics temperature too low
diagnostic message.

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Factory setting

Out of specification (S)Additional information

Selection
For a detailed description of the event categories available for selection: →  131

Event category 834 (Process temperature too high)

NavigationDescription

Use this option to select a category for the 834 Process temperature too high diagnostic
message.

134

Expert → Communication → Diag. config. → Event category 834

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Factory setting

Out of specification (S)Additional information

Selection
For a detailed description of the event categories available for selection: →  131

Event category 835 (Process temperature too low)

Navigation

Expert → Communication → Diag. config. → Event category 835

Description

Use this option to select a category for the 835 Process temperature too low diagnostic
message.

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Factory setting

Out of specification (S)Additional information

Selection
For a detailed description of the event categories available for selection: →  131

3.6.4"Web server" submenu

Navigation

 Expert → Communication → Web server‣ Web serverEndress+Hauser

Web server language→  136

MAC address→  136

DHCP client→  137

IP address→  137

Subnet mask

→  137135

Description of device parameters

Proline Teqwave M 300 HART

Default gateway→  138

Web server functionality→  138

Login page→  138

Web server language
Navigation

 Expert → Communication → Web server → Webserv.language

Description

Use this function to select the language configured for the Web server.

Selection• English• Deutsch• Français• Español• Italiano• Nederlands• Portuguesa• Polski• русский язык (Russian)• Svenska• Türkçe• 中文 (Chinese)• 日本語 (Japanese)• 한국어 (Korean)• čeština (Czech)

Factory setting

English

MAC address
Navigation

 Expert → Communication → Web server → MAC Address

Description

Displays the MAC 4) address of the measuring device.

User interface

Unique 12-digit character string comprising letters and numbers

Factory setting

Each measuring device is given an individual address.

Additional information

ExampleFor the display format
00:07:05:10:01:5F

4)

136

Media Access Control

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

DHCP clientNavigation

 Expert → Communication → Web server → DHCP client

Description

Use this function to activate and deactivate the DHCP client functionality.

Selection

• Off
• On

Factory setting

On

Additional information

EffectIf the DHCP client functionality of the web server is selected, the IP address (→  137),
Subnet mask (→  137) and Default gateway (→  138) are set automatically.
• Identification is via the MAC address of the measuring device.
• The IP address (→  137) in the IP address parameter (→  137) is ignored aslong as the DHCP client parameter (→  137) is active. This is also the case, inparticular, if the DHCP server cannot be reached. The IP address (→  137) in theparameter of the same name is only used if the DHCP client parameter (→  137)
is inactive.

IP addressNavigation

 Expert → Communication → Web server → IP address

Description

Display or enter the IP address of the Web server integrated in the measuring device.

User entry

4 octet: 0 to 255 (in the particular octet)

Factory setting

192.168.1.212

Subnet mask

Navigation

 Expert → Communication → Web server → Subnet mask

Description

Display or enter the subnet mask.

User entry

4 octet: 0 to 255 (in the particular octet)

Factory setting

255.255.255.0

Endress+Hauser

137

Description of device parameters

Proline Teqwave M 300 HART

Default gateway

Navigation

 Expert → Communication → Web server → Default gateway

Description

Display or enter the Default gateway (→  138).

User entry

4 octet: 0 to 255 (in the particular octet)

Factory setting

0.0.0.0

Web server functionality



Navigation

 Expert → Communication → Web server → Webserver funct.

Description

Use this function to switch the Web server on and off.

Selection• Off• HTML Off• On

Factory setting

On

Additional information

Description
Once disabled, the Web server functionality can only be enabled again via the local
display, the FieldCare operating tool or the DeviceCare operating tool.
Selection
OptionDescription

Off• The Web server is completely disabled.
• Port 80 is locked.

HTML OffThe HTML version of the Web server is not available.

On• The complete Web server functionality is available.
• JavaScript is used.
• The password is transferred in an encrypted state.
• Any change to the password is also transferred in an encrypted state.

Login page

Navigation

 Expert → Communication → Web server → Login page

Description

Use this function to select the format of the login page.

Selection• Without header• With header

138

Endress+Hauser

Proline Teqwave M 300 HART

Factory setting

Description of device parameters

With header

3.6.5"WLAN settings" wizard

Navigation

 Expert → Communication → WLAN settings‣ WLAN settings

Endress+Hauser

WLAN→  140WLAN mode→  140

SSID name→  140

Network security→  141

Security identification→  141

User name→  141

WLAN password→  142WLAN IP address→  142WLAN MAC address→  142WLAN subnet mask

→  143WLAN MAC address→  142WLAN passphrase→  143WLAN MAC address→  142

Assign SSID name→  143

SSID name→  144

2.4 GHz WLAN channel→  144

Select antenna→  144

Connection state→  144

Received signal strength→  145139

Description of device parameters

Proline Teqwave M 300 HARTWLAN IP address→  142

Gateway IP address→  145

IP address domain name server→  145WLAN

Navigation

 Expert → Communication → WLAN settings → WLAN

Description

Use this function to enable and disable the WLAN connection.

Selection• Disable• Enable

Factory setting

EnableWLAN modeNavigation

 Expert → Communication → WLAN settings → WLAN mode

Description

Use this function to select the WLAN mode.

Selection• WLAN access point• WLAN Client

Factory setting

WLAN access point

SSID name

Navigation

 Expert → Communication → WLAN settings → SSID name

Prerequisite

The client is activated.

Description

Use this function to enter the user-defined SSID name (max. 32 characters) of the WLAN
network.

User entry–

Factory setting–

140

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Network security

Navigation

 Expert → Communication → WLAN settings → Network security

Description

Use this function to select the type of security for the WLAN interface.

Selection• Unsecured• WPA2-PSK• EAP-PEAP with MSCHAPv2 *• EAP-PEAP MSCHAPv2 no server authentic. *• EAP-TLS *Factory setting

WPA2-PSK

Additional information

Selection• Unsecured
Access the WLAN connection without identification.
• WPA2-PSK
Access the WLAN connection with a network key.
• EAP-PEAP with MSCHAPv2
Access the WLAN connection with a password-based authentication protocol.
• EAP-PEAP MSCHAPv2 no server authentic.
Access the WLAN connection with a password-based protocol without server
authentication.
• EAP-TLS
Access the WLAN connection with a certificate-based, two-way authentication of theclient and network.

Security identification
Navigation

 Expert → Communication → WLAN settings → Sec. identific.

Description

Use this function to select the security settings (download via the menu: DataManagement > Security > Download WLAN).

User interface• Trusted issuer certificate• Device certificate• Device private key

User name

Navigation

 Expert → Communication → WLAN settings → User name

Description

Use this function to enter the username of the WLAN network.

User entry–*

Visibility depends on order options or device settings

Endress+Hauser

141

Description of device parameters

Factory setting

Proline Teqwave M 300 HART–WLAN passwordNavigation

 Expert → Communication → WLAN settings → WLAN password

Description

Use this function to enter the WLAN password for the WLAN network.

User entry–

Factory setting–

WLAN IP addressNavigation

 Expert → Communication → WLAN settings → WLAN IP address

Description

Use this function to enter the IP address of the measuring device's WLAN connection.

User entry

4 octet: 0 to 255 (in the particular octet)

Factory setting

192.168.1.212

WLAN MAC address
Navigation

 Expert → Communication → WLAN settings → WLAN MAC address

Description

Displays the MAC 5) address of the measuring device.

User interface

Unique 12-digit character string comprising letters and numbers

Factory setting

Each measuring device is given an individual address.

Additional information

ExampleFor the display format
00:07:05:10:01:5F

5)

142

Media Access Control

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

WLAN subnet mask
Navigation



Expert → Communication → WLAN settings → WLAN subnet mask

Expert → Communication → WLAN settings → WLAN subnet mask

 Expert → Communication → WLAN settings → WLAN subnet mask
Description

Use this function to enter the subnet mask.

User entry

4 octet: 0 to 255 (in the particular octet)

Factory setting

255.255.255.0

WLAN passphraseNavigation

 Expert → Communication → WLAN settings → WLAN passphrase

Prerequisite

The WPA2-PSK option is selected in the Security type parameter (→  141).

Description

Use this function to enter the network key.

User entry

8 to 32-digit character string comprising numbers, letters and special characters (without
spaces)

Factory setting

Serial number of the measuring device (e.g. L100A802000)

Assign SSID name

Navigation

 Expert → Communication → WLAN settings → Assign SSID name

Description

Use this function to select which name is used for the SSID 6).

Selection• Device tag• User-defined

Factory settingUser-defined

Additional information

Selection• Device tag
The device tag name is used as the SSID.
• User-defined
A user-defined name is used as the SSID.

6)

Service Set Identifier

Endress+Hauser

143

Description of device parameters

Proline Teqwave M 300 HART

SSID name

Navigation

 Expert → Communication → WLAN settings → SSID name

Prerequisite• The User-defined option is selected in the Assign SSID name parameter (→  143).
• The WLAN access point option is selected in the WLAN mode parameter (→  140).

Description

Use this function to enter a user-defined SSID name.

User entry

Max. 32-digit character string comprising numbers, letters and special characters

Factory setting

2.4 GHz WLAN channel

Navigation

 Expert → Communication → WLAN settings → WLAN channel

Description

Use this function to enter the 2.4 GHz WLAN channel.

User entry

1 to 11

Factory setting

6

Additional information

Description• It is only necessary to enter a 2.4 GHz WLAN channel if multiple WLAN devices arein use.
• If just one measuring device is in use, it is recommended to keep the factory setting.

Select antennaNavigation

 Expert → Communication → WLAN settings → Select antenna

Description

Use this function to select whether the external or internal antenna is used for reception.

Selection• External antenna• Internal antenna

Factory setting

Internal antenna

Connection state
NavigationDescription

The connection status is displayed.

144

Expert → Communication → WLAN settings → Connection state

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

User interface• Connected• Not connected

Factory setting

Not connected

Received signal strength
Navigation

 Expert → Communication → WLAN settings → Rec.sig.strength

Description

Displays the signal strength received.

User interface• Low• Medium• HighFactory setting

High

Gateway IP address
Navigation

 Expert → Communication → WLAN settings → Gateway IP addr.

Description

Use this function to enter the IP address of the gateway.

User interface

Character string comprising numbers, letters and special characters

Factory setting

192.168.1.212

IP address domain name server
Navigation

 Expert → Communication → WLAN settings → IP address DNS

Description

Use this function to enter the IP address of the domain name server.

User interface

Character string comprising numbers, letters and special characters

Factory setting

192.168.1.212

Endress+Hauser

145

Description of device parameters

Proline Teqwave M 300 HART

3.7

"Application" submenu

Navigation

 Expert → Application‣ Application
Reset all totalizers→  146‣ Totalizer 1→  146

Reset all totalizers
Navigation

 Expert → Application → Reset all tot.

Description

Use this function to reset all totalizers to the value 0 and restart the totaling process. Thisdeletes all the previously aggregated flow values.

Selection• Cancel• Reset + totalize

Factory setting

CancelAdditional information

Selection
Options

Description

CancelNo action is executed and the user exits the parameter.

Reset + totalize

Resets the totalizer to 0 and restarts the totaling process. The previously
aggregated load quantity is thus deleted.

3.7.1"Totalizer 1 to n" submenu

Navigation

 Expert → Application → Totalizer 1 to n

‣ Totalizer 1

146

Assign process variable 1→  147

Process variable unit 1→  147

Totalizer 1 operation mode→  148

Totalizer 1 control→  148

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Preset value 1→  149

Totalizer 1 failure behavior→  149

Assign process variable 1Navigation

 Expert → Application → Totalizer 1 → AssignVariab. 1

Prerequisite

The Load rate option is only available if the volume flow of the medium is read in via theCurrent input 1 to n (→  49) or the fieldbus.

Description

Use this function to select a process variable for the Totalizer 1 to n.

Selection• Off• Load rate *Factory setting

Off

Additional information

Description
If the option selected is changed, the device resets the totalizer to 0.
Options
If the Off option is selected, only the Assign process variable parameter (→  147) is
still displayed in the Totalizer 1 to n submenu. All other parameters in the submenu arehidden.

Process variable unit 1Navigation

 Expert → Application → Totalizer 1 → VariableUnit 1

Prerequisite

A process variable is selected in the Assign process variable parameter (→  147) of the
Totalizer 1 to n submenu.

Description

Use this function to select the process variable unit for the Totalizer 1 to n (→  146).

Selection

SI units• kg•t

US units• oz• lb• STonor*

Visibility depends on order options or device settings

Endress+Hauser

147

Description of device parameters

Proline Teqwave M 300 HART

Other units
None **

Visibility depends on order options or device settings

Factory setting

Depends on country

Additional information

Description
The unit is selected separately for the totalizer and is independent of the option
selected in the System units submenu (→  54).
Options
The selection is dependent on the process variable selected in the Assign process variable
parameter (→  147).

Totalizer 1 operation modeNavigation

 Expert → Application → Totalizer 1 → Operat. mode 1

Prerequisite

A process variable is selected in the Assign process variable parameter (→  147) of the
Totalizer 1 to n submenu.

Description

Use this function to select how the totalizer summates the flow.

Selection• Net• Forward• Reverse

Factory setting

Net

Additional information

Selection• Net flow total
Flow values in the forward and reverse flow direction are totalized and balanced against
one another. Net flow is registered in the flow direction.
• Forward flow total
Only the flow in the forward flow direction is totalized.
• Reverse flow total
Only the flow in the reverse flow direction is totalized (= reverse flow quantity).

Totalizer 1 control
Navigation

 Expert → Application → Totalizer 1 → Tot. 1 control

Prerequisite

A process variable is selected in the Assign process variable parameter (→  147) of the
Totalizer 1 to n submenu.

Description

Use this function to select the control of totalizer value 1-3.

148

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Selection• Totalize• Reset + hold
• Preset + hold
• Reset + totalize• Preset + totalize• HoldFactory setting

Totalize

Additional information

Selection
Options

Description

Totalize

The totalizer is started or continues running.

Reset + holdPreset + hold

The totaling process is stopped and the totalizer is reset to 0.
1)

The totaling process is stopped and the totalizer is set to its defined start value
from the Preset value parameter.

Reset + totalize
Preset + totalize
Hold
1)

The totalizer is reset to 0 and the totaling process is restarted.
1)

The totalizer is set to the defined start value in the Preset value parameter and thetotaling process is restarted.
Totalizing is stopped.

Visible depending on the order options or device settings

Preset value 1
Navigation

 Expert → Application → Totalizer 1 → Preset value 1

Prerequisite

A process variable is selected in the Assign process variable parameter (→  147) of the
Totalizer 1 to n submenu.

Description

Use this function to enter a start value for the Totalizer 1 to n.

User entry

Signed floating-point number

Factory setting

0 kg

Additional information

User entry
The unit of the selected process variable is defined in the Unit totalizer parameter(→  147) for the totalizer.

Totalizer 1 failure behavior

Navigation

 Expert → Application → Totalizer 1 → FailureBehav. 1

Prerequisite

A process variable is selected in the Assign process variable parameter (→  147) of the
Totalizer 1 to n submenu.

Description

Use this function to select how a totalizer behaves in the event of a device alarm.

Endress+Hauser

149

Description of device parameters

Proline Teqwave M 300 HART

Selection• Hold• Continue• Last valid value + continue

Factory setting

Hold

Additional information

Description
This setting does not affect the failsafe mode of other totalizers and the outputs. Thisis specified in separate parameters.
Selection• Stop
The totalizer is stopped in the event of a device alarm.
• Actual value
The totalizer continues to count based on the actual (current) measured value; the device
alarm is ignored.
• Last valid value
The totalizer continues to count based on the last valid measured value before the device
alarm occurred.

3.8"Diagnostics" submenu

Navigation

 Expert → Diagnostics‣ Diagnostics
Actual diagnostics→  151

Previous diagnostics→  152

Operating time from restart→  152

Operating time→  153‣ Diagnostic list

→  153‣ Event logbook→  155‣ Device information→  157‣ Main electronic module + I/O→  161‣ Sensor electronic module (ISEM)→  162‣ I/O module 2→  163‣ I/O module 3→  164

module 1

150

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters‣ Display module→  165‣ Data logging→  166‣ Min/max values→  174‣ Heartbeat Technology→  178‣ Simulation→  179

Actual diagnostics
Navigation

 Expert → Diagnostics → Actual diagnos.

Prerequisite

A diagnostic event has occurred.

Description

Displays the current diagnostic message. If two or more messages occur simultaneously,
the message with the highest priority is shown on the display.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

Additional information

Display
Additional pending diagnostic messages can be viewed in the Diagnostic list
submenu (→  153).
Via the local display: the time stamp and corrective measures referring to the cause of
the diagnostic message can be accessed via the  key.
ExampleFor the display format:
F271 Main electronics failure

Timestamp
Navigation

Expert → Diagnostics → Timestamp

Description

Displays the operating time when the current diagnostic message occurred.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

Display
The diagnostic message can be viewed via the Actual diagnostics parameter(→  151).
ExampleFor the display format:
24d12h13m00s

Endress+Hauser

151

Description of device parameters

Proline Teqwave M 300 HART

Previous diagnostics
Navigation

 Expert → Diagnostics → Prev.diagnostics

Prerequisite

Two diagnostic events have already occurred.

Description

Displays the diagnostic message that occurred before the current message.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

Additional information

Display
Via the local display: the time stamp and corrective measures referring to the cause of
the diagnostic message can be accessed via the  key.
ExampleFor the display format:
F271 Main electronics failure

Timestamp
Navigation

Expert → Diagnostics → Timestamp

Description

Displays the operating time when the last diagnostic message before the current message
occurred.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

Display
The diagnostic message can be viewed via the Previous diagnostics parameter(→  152).
ExampleFor the display format:
24d12h13m00s

Operating time from restart
Navigation

 Expert → Diagnostics → Time fr. restart

Description

Use this function to display the time the device has been in operation since the last device
restart.

User interface

Days (d), hours (h), minutes (m) and seconds (s)152

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Operating time
Navigation

 Expert → Diagnostics → Operating time

Description

Displays the length of time the device has been in operation.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

Indication
Maximum number of days: 9 999(corresponds to approx. 27 years and 5 months)

3.8.1"Diagnostic list" submenu

Navigation

 Expert → Diagnostics → Diagnostic list

‣ Diagnostic list
Diagnostics 1→  153

Diagnostics 2→  154

Diagnostics 3→  154

Diagnostics 4→  154

Diagnostics 5→  155

Diagnostics 1
Navigation

 Expert → Diagnostics → Diagnostic list → Diagnostics 1

Description

Displays the current diagnostics message with the highest priority.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

Additional information

Display
Via the local display: the time stamp and corrective measures referring to the cause of
the diagnostic message can be accessed via the  key.
Examples
For the display format:• F271 Main electronics failure• F276 I/O module failure

Endress+Hauser

153

Description of device parameters

Proline Teqwave M 300 HART

Diagnostics 2
Navigation

 Expert → Diagnostics → Diagnostic list → Diagnostics 2

Description

Displays the current diagnostics message with the second-highest priority.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

Additional information

Display
Via the local display: the time stamp and corrective measures referring to the cause of
the diagnostic message can be accessed via the  key.
Examples
For the display format:• F271 Main electronics failure• F276 I/O module failure

Diagnostics 3
Navigation

 Expert → Diagnostics → Diagnostic list → Diagnostics 3

Description

Displays the current diagnostics message with the third-highest priority.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

Additional information

Display
Via the local display: the time stamp and corrective measures referring to the cause of
the diagnostic message can be accessed via the  key.
Examples
For the display format:• F271 Main electronics failure• F276 I/O module failure

Diagnostics 4
Navigation

 Expert → Diagnostics → Diagnostic list → Diagnostics 4

Description

Displays the current diagnostics message with the fourth-highest priority.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

154

Endress+Hauser

Proline Teqwave M 300 HART

Additional information

Description of device parameters

Display
Via the local display: the time stamp and corrective measures referring to the cause of
the diagnostic message can be accessed via the  key.
Examples
For the display format:• F271 Main electronics failure• F276 I/O module failure

Diagnostics 5
Navigation

 Expert → Diagnostics → Diagnostic list → Diagnostics 5

Description

Displays the current diagnostics message with the fifth-highest priority.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

Additional information

Display
Via the local display: the time stamp and corrective measures referring to the cause of
the diagnostic message can be accessed via the  key.
Examples
For the display format:• F271 Main electronics failure• F276 I/O module failure

3.8.2"Event logbook" submenu

Viewing event messages
Event messages are displayed in chronological order. The event history includes both
diagnostic events and information events. The symbol in front of the timestamp indicates
whether the event has started or ended.
Navigation

Expert → Diagnostics → Event logbook‣ Event logbook
Filter options

Endress+Hauser→  156

155

Description of device parameters

Proline Teqwave M 300 HART

Filter options

Navigation

Expert → Diagnostics → Event logbook → Filter options

Description

Use this function to select the category whose event messages are displayed in the event
list of the local display.

Selection• All• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• Information (I)Factory setting

All

Additional information

Description
The status signals are categorized in accordance with VDI/VDE 2650 and NAMUR
Recommendation NE 107:
• F = Failure• C = Function Check• S = Out of Specification• M = Maintenance Required"Event list" submenu
The Event list submenu is only displayed if operating via the local display.
If operating via the FieldCare operating tool, the event list can be read out with aseparate FieldCare module.
If operating via the Web browser, the event messages can be found directly in the
Event logbook submenu.
Navigation

 Expert → Diagnostics → Event logbook → Event list

▸ Event list
Event list

→  156

Event list
NavigationDescription

Displays the history of event messages of the category selected in the Filter options
parameter (→  156).

156

Expert → Diagnostics → Event logbook → Event list

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

User interface• For a "Category I" event message
Information event, short message, symbol for event recording and operating time whenerror occurred• For a "Category F, C, S, M" event message (status signal)
Diagnostics code, short message, symbol for event recording and operating time whenerror occurred

Additional information

Description
A maximum of 20 event messages are displayed in chronological order.
If the Extended HistoROM application package (order option) is enabled in the device, theevent list can contain up to 100 entries .
The following symbols indicate whether an event has occurred or has ended:• : Occurrence of the event• : End of the event
Examples
For the display format:• I1091 Configuration modified
 24d12h13m00s• F271 Main electronics failure 01d04h12min30s
HistoROM
A HistoROM is a "non-volatile" device memory in the form of an EEPROM.

3.8.3
Navigation"Device information" submenu
 Expert → Diagnostics → Device info

‣ Device information

Endress+Hauser

Device tag→  158

Serial number→  158

Firmware version→  159

Device name→  159

Order code→  159

Extended order code 1→  159

Extended order code 2→  160

Extended order code 3→  160

157

Description of device parameters

Proline Teqwave M 300 HART

Configuration counter→  160

ENP version→  160

Device tag
Navigation

 Expert → Diagnostics → Device info → Device tag

Description

Displays a unique name for the measuring point so it can be identified quickly within theplant. It is displayed in the header.

User interface

Max. 32 characters, such as letters, numbers or special characters (e.g. @, %, /).

Factory setting

Teqwave M

Additional information

User interface
1

XXXXXXXXX

A0029422

1

Position of the header text on the display

The number of characters displayed depends on the characters used.

Serial number
Navigation

 Expert → Diagnostics → Device info → Serial number

Description

Displays the serial number of the measuring device.
The number can be found on the nameplate of the sensor and transmitter.

User interface

Max. 11-digit character string comprising letters and numbers.

Additional information

Description
Uses of the serial number• To identify the measuring device quickly, e.g. when contacting Endress+Hauser.
• To obtain specific information on the measuring device using the Device Viewer:
www.endress.com/deviceviewer

158

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Firmware version
Navigation

 Expert → Diagnostics → Device info → Firmware version

Description

Displays the device firmware version installed.

User interface

Character string in the format xx.yy.zz

Additional information

Display
The Firmware version is also located:• On the title page of the Operating instructions• On the transmitter nameplate

Device name
Navigation

 Expert → Diagnostics → Device info → Device name

Description

Displays the name of the transmitter. It can also be found on the nameplate of thetransmitter.

User interface

Character string comprising numbers, letters and special characters

Order code

Navigation

 Expert → Diagnostics → Device info → Order code

Description

Displays the device order code.

User interface

Character string composed of letters, numbers and certain punctuation marks (e.g. /).

Additional information

Description
The order code is generated from the extended order code through a process of reversible
transformation. The extended order code indicates the attributes for all the device featuresin the product structure. The device features are not directly readable from the order code.
Uses of the order code
• To order an identical spare device.
• To identify the device quickly and easily, e.g. when contacting Endress+Hauser.

Extended order code 1Navigation

 Expert → Diagnostics → Device info → Ext. order cd. 1

Description

Displays the first part of the extended order code.

Endress+Hauser

159

Description of device parameters

Proline Teqwave M 300 HARTOn account of length restrictions, the extended order code is split into a maximum of 3
parameters.
User interface

Character string

Additional information

Description
The extended order code indicates the version of all the features of the product structure
for the measuring device and thus uniquely identifies the measuring device.

Extended order code 2Navigation

 Expert → Diagnostics → Device info → Ext. order cd. 2

Description

Displays the second part of the extended order code.

User interface

Character string

Additional information

For additional information, see Extended order code 1 parameter (→  159)

Extended order code 3

Navigation

 Expert → Diagnostics → Device info → Ext. order cd. 3

Description

Displays the third part of the extended order code.

User interface

Character string

Additional information

For additional information, see Extended order code 1 parameter (→  159)

Configuration counter
Navigation

 Expert → Diagnostics → Device info → Config. counter

Description

Displays the number of parameter modifications for the device. When the user changes aparameter setting, this counter is incremented.

User interface

0 to 65 535

ENP version
Navigation

 Expert → Diagnostics → Device info → ENP version

Description

Displays the version of the electronic nameplate.

160

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

User interface

Character string

Factory setting

2.02.00

Additional information

Description
This electronic nameplate stores a data record for device identification that includes moredata than the nameplates attached to the outside of the device.

3.8.4"Main electronic module + I/O module 1" submenu

Navigation

 Expert → Diagnostics → Main elec.+I/O1‣ Main electronic module + I/O
module 1

Firmware version→  161

Build no. software→  161

Bootloader revision→  162

Firmware version
Navigation

 Expert → Diagnostics → Main elec.+I/O1 → Firmware version

Description

Use this function to display the software revision of the module.

User interface

Positive integer

Build no. software
Navigation

 Expert → Diagnostics → Main elec.+I/O1 → Build no. softw.

Description

Use this function to display the software build number of the module.

User interface

Positive integer

Endress+Hauser

161

Description of device parameters

Proline Teqwave M 300 HART

Bootloader revision
Navigation

 Expert → Diagnostics → Main elec.+I/O1 → Bootloader rev.

Description

Use this function to display the bootloader revision of the software.

User interface

Positive integer

3.8.5"Sensor electronic module (ISEM)" submenu

Navigation

 Expert → Diagnostics → Sens. electronic‣ Sensor electronic module (ISEM)
Firmware version→  162

Build no. software→  162

Bootloader revision→  163

Firmware version
Navigation

 Expert → Diagnostics → Sens. electronic → Firmware version

Description

Use this function to display the software revision of the module.

User interface

Positive integer

Build no. software
Navigation

 Expert → Diagnostics → Sens. electronic → Build no. softw.

Description

Use this function to display the software build number of the module.

User interface

Positive integer

162

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Bootloader revision
Navigation

 Expert → Diagnostics → Sens. electronic → Bootloader rev.

Description

Use this function to display the bootloader revision of the software.

User interface

Positive integer

3.8.6"I/O module 2" submenu

Navigation

 Expert → Diagnostics → I/O module 2‣ I/O module 2
I/O module 2 terminal numbers→  163

Firmware version→  163

Build no. software→  164

Bootloader revision→  164

I/O module 2 terminal numbers
Navigation

 Expert → Diagnostics → I/O module 2 → I/O 2 terminals

Description

Displays the terminal numbers used by the I/O module.

User interface• Not used
• 26-27 (I/O 1)• 24-25 (I/O 2)• 22-23 (I/O 3)

Firmware version
Navigation

 Expert → Diagnostics → I/O module 2 → Firmware version

Description

Use this function to display the software revision of the module.

User interface

Positive integer

Endress+Hauser

163

Description of device parameters

Proline Teqwave M 300 HART

Build no. software
Navigation

 Expert → Diagnostics → I/O module 2 → Build no. softw.

Description

Use this function to display the software build number of the module.

User interface

Positive integer

Bootloader revision
Navigation

 Expert → Diagnostics → I/O module 2 → Bootloader rev.

Description

Use this function to display the bootloader revision of the software.

User interface

Positive integer

3.8.7"I/O module 3" submenu

Navigation

 Expert → Diagnostics → I/O module 3‣ I/O module 3
I/O module 3 terminal numbers→  164

Firmware version→  165

Build no. software→  165

Bootloader revision→  165

I/O module 3 terminal numbers
Navigation

 Expert → Diagnostics → I/O module 3 → I/O 3 terminals

Description

Displays the terminal numbers used by the I/O module.

User interface• Not used
• 26-27 (I/O 1)• 24-25 (I/O 2)• 22-23 (I/O 3)

164

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Firmware version
Navigation

 Expert → Diagnostics → I/O module 3 → Firmware version

Description

Use this function to display the software revision of the module.

User interface

Positive integer

Build no. software
Navigation

 Expert → Diagnostics → I/O module 3 → Build no. softw.

Description

Use this function to display the software build number of the module.

User interface

Positive integer

Bootloader revision
Navigation

 Expert → Diagnostics → I/O module 3 → Bootloader rev.

Description

Use this function to display the bootloader revision of the software.

User interface

Positive integer

3.8.8"Display module" submenu

Navigation

 Expert → Diagnostics → Display module‣ Display module

Endress+Hauser

Firmware version→  166

Build no. software→  166

Bootloader revision→  166165

Description of device parameters

Proline Teqwave M 300 HART

Firmware version
Navigation

 Expert → Diagnostics → Display module → Firmware version

Description

Use this function to display the software revision of the module.

User interface

Positive integer

Build no. software
Navigation

 Expert → Diagnostics → Display module → Build no. softw.

Description

Use this function to display the software build number of the module.

User interface

Positive integer

Bootloader revision
Navigation

 Expert → Diagnostics → Display module → Bootloader rev.

Description

Use this function to display the bootloader revision of the software.

User interface

Positive integer

3.8.9"Data logging" submenu

Navigation

 Expert → Diagnostics → Data logging‣ Data logging

166

Assign channel 1→  167

Assign channel 2→  168

Assign channel 3→  168

Assign channel 4→  168

Logging interval→  169

Clear logging data

→  169

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Data logging→  170

Logging delay→  170

Data logging control→  170

Data logging status→  171

Entire logging duration→  171

Assign channel 1Navigation

 Expert → Diagnostics → Data logging → Assign chan. 1

Prerequisite• The Load rate option is only available if the volume flow of the medium is read in via theCurrent input 1 to n (→  49) or the fieldbus.
• The Extended HistoROM application package is available.
The software options currently enabled are displayed in the Software option
overview parameter (→  45).

Description

Use this function to assign a process variable to the data logging channel.

Selection• Off• Total solids• Temperature• Electronics temperature• Conductivity• Corrected conductivity• Load rate *• Current output 1 *• Current output 2 *• Current output 3 *• Current output 4 *Factory setting

Off

Additional information

Description
A total of 1000 measured values can be logged. This means:• 1000 data points if 1 logging channel is used
• 500 data points if 2 logging channels are used
• 333 data points if 3 logging channels are used
• 250 data points if 4 logging channels are used
Once the maximum number of data points is reached, the oldest data points in the data log
are cyclically overwritten in such a way that the last 1000, 500, 333 or 250 measured
values are always in the log (ring memory principle).
The log contents are cleared if the option selected is changed.

*

Visibility depends on order options or device settings

Endress+Hauser

167

Description of device parameters

Proline Teqwave M 300 HART

Assign channel 2

Navigation

 Expert → Diagnostics → Data logging → Assign chan. 2

Prerequisite• The Load rate option is only available if the volume flow of the medium is read in via theCurrent input 1 to n (→  49) or the fieldbus.
• The Extended HistoROM application package is available.
The software options currently enabled are displayed in the Software option
overview parameter (→  45).

Description

Use this function to assign a process variable to the logging channel.

Selection

For the picklist, see Assign channel 1 parameter (→  167)

Factory setting

Off

Assign channel 3

Navigation

 Expert → Diagnostics → Data logging → Assign chan. 3

Prerequisite• The Load rate option is only available if the volume flow of the medium is read in via theCurrent input 1 to n (→  49) or the fieldbus.
• The Extended HistoROM application package is available.
The software options currently enabled are displayed in the Software option
overview parameter (→  45).

Description

Use this function to assign a process variable to the logging channel.

Selection

For the picklist, see Assign channel 1 parameter (→  167)

Factory setting

Off

Assign channel 4Navigation

 Expert → Diagnostics → Data logging → Assign chan. 4

Prerequisite• The Load rate option is only available if the volume flow of the medium is read in via theCurrent input 1 to n (→  49) or the fieldbus.
• The Extended HistoROM application package is available.
The software options currently enabled are displayed in the Software option
overview parameter (→  45).

Description

Use this function to assign a process variable to the logging channel.

Selection

For the picklist, see Assign channel 1 parameter (→  167)

Factory setting

Off

168

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Logging interval

Navigation

 Expert → Diagnostics → Data logging → Logging interval

Prerequisite

The Extended HistoROM application package is available.
The software options currently enabled are displayed in the Software option
overview parameter (→  45).

Description

Use this function to enter the logging interval Tlog for data logging.

User entry

0.1 to 3 600.0 s

Factory setting

1.0 s

Additional information

Description
This defines the interval between the individual data points in the data log, and thus the
maximum loggable process time Tlog:• If 1 logging channel is used: Tlog = 1000 × tlog
• If 2 logging channels are used: Tlog = 500 × tlog
• If 3 logging channels are used: Tlog = 333 × tlog
• If 4 logging channels are used: Tlog = 250 × tlog
Once this time elapses, the oldest data points in the data log are cyclically overwritten suchthat a time of Tlog always remains in the memory (ring memory principle).
The log contents are cleared if the length of the logging interval is changed.
ExampleIf 1 logging channel is used:• Tlog = 1000 × 1 s = 1 000 s ≈15 min• Tlog = 1000 × 10 s = 10 000 s ≈ 3 h• Tlog = 1000 × 80 s = 80 000 s ≈ 1 d• Tlog = 1000 × 3 600 s = 3 600 000 s ≈ 41 d

Clear logging dataNavigation

 Expert → Diagnostics → Data logging → Clear logging

Prerequisite

The Extended HistoROM application package is available.
The software options currently enabled are displayed in the Software option
overview parameter (→  45).

Description

Use this function to clear the entire logging data.

Selection• Cancel• Clear data

Factory setting

CancelEndress+Hauser

169

Description of device parameters

Additional information

Proline Teqwave M 300 HART

Selection• Cancel
The data is not cleared. All the data is retained.
• Clear data
The logging data is cleared. The logging process starts from the beginning.

Data logging

Navigation

 Expert → Diagnostics → Data logging → Data logging

Description

Use this function to select the data logging method.

Selection• Overwriting• Not overwriting

Factory setting

Overwriting

Additional information

Selection• Overwriting
The device memory applies the FIFO principle.
• Not overwriting
Data logging is canceled if the measured value memory is full (single shot).

Logging delay

Navigation

 Expert → Diagnostics → Data logging → Logging delay

Prerequisite

In the Data logging parameter (→  170), the Not overwriting option is selected.

Description

Use this function to enter the time delay for measured value logging.

User entry

0 to 999 h

Factory setting

0hAdditional information

Description
Once data logging has been started with the Data logging control parameter (→  170),
the device does not save any data for the duration of the delay time entered.

Data logging control



Navigation

 Expert → Diagnostics → Data logging → Data log.control

Prerequisite

In the Data logging parameter (→  170), the Not overwriting option is selected.

170

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Description

Use this function to start and stop measured value logging.

Selection• None• Delete + start• Stop

Factory setting

None

Additional information

Selection• NoneInitial measured value logging status.
• Delete + start
All the measured values recorded for all the channels are deleted and measured value
logging starts again.
• StopMeasured value logging is stopped.

Data logging status
Navigation

 Expert → Diagnostics → Data logging → Data log. status

Prerequisite

In the Data logging parameter (→  170), the Not overwriting option is selected.

Description

Displays the measured value logging status.

User interface• Done• Delay active• Active• Stopped

Factory setting

Done

Additional information

Selection• DoneMeasured value logging has been performed and completed successfully.
• Delay active
Measured value logging has been started but the logging interval has not yet elapsed.
• Active
The logging interval has elapsed and measured value logging is active.
• Stopped
Measured value logging is stopped.

Entire logging duration
Navigation

 Expert → Diagnostics → Data logging → Logging duration

Prerequisite

In the Data logging parameter (→  170), the Not overwriting option is selected.

Description

Displays the total logging duration.

Endress+Hauser

171

Description of device parameters

Proline Teqwave M 300 HART

User interface

Positive floating-point number

Factory setting

0s"Display channel 1" submenu
Navigation

Expert → Diagnostics → Data logging → Displ.channel 1▸ Display channel 1
Display channel 1→  172

Display channel 1
Navigation

Expert → Diagnostics → Data logging → Displ.channel 1

Prerequisite

The Extended HistoROM application package is available.
The software options currently enabled are displayed in the Software option
overview parameter (→  45).

Description

Displays the measured value trend for the logging channel in the form of a chart.

Additional information

Description• x-axis: depending on the number of channels selected displays 250 to 1000 measured
values of a process variable.
• y-axis: displays the approximate measured value span and constantly adapts this to theongoing measurement.

"Display channel 2" submenu
Navigation

Expert → Diagnostics → Data logging → Displ.channel 2▸ Display channel 2
Display channel 2172→  173

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Display channel 2
Navigation

Expert → Diagnostics → Data logging → Displ.channel 2

Prerequisite

A process variable is specified in the Assign channel 2 parameter.

Description

See the Display channel 1 parameter →  172"Display channel 3" submenu
Navigation

Expert → Diagnostics → Data logging → Displ.channel 3▸ Display channel 3
Display channel 3→  173

Display channel 3
Navigation

Expert → Diagnostics → Data logging → Displ.channel 3

Prerequisite

A process variable is specified in the Assign channel 3 parameter.

Description

See the Display channel 1 parameter →  172"Display channel 4" submenu
Navigation

Expert → Diagnostics → Data logging → Displ.channel 4▸ Display channel 4
Display channel 4→  173

Display channel 4
Navigation

Prerequisite

A process variable is specified in the Assign channel 4 parameter.

Endress+Hauser

Expert → Diagnostics → Data logging → Displ.channel 4

173

Description of device parameters

Description

Proline Teqwave M 300 HART

See the Display channel 1 parameter →  172

3.8.10"Min/max values" submenu

Navigation

 Expert → Diagnostics → Min/max val.

‣ Min/max values‣ Electronics temperature→  174‣ Main electronics temperature→  175‣ Sensor electronics temperature→  176‣ Medium temperature→  176‣ Total solids→  177‣ Conductivity→  178(ISEM)"Electronics temperature" submenu
Navigation

 Expert → Diagnostics → Min/max val. → Electronics temp

‣ Electronics temperature
Minimum value→  174

Maximum value→  175

Minimum value
Navigation

 Expert → Diagnostics → Min/max val. → Electronics temp → Minimum value

Description

Shows the lowest electronics temperature measured to date.

User interface

Positive floating-point number

174

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Maximum value
Navigation

 Expert → Diagnostics → Min/max val. → Electronics temp → Maximum value

Description

Shows the highest electronics temperature measured to date.

User interface

Positive floating-point number"Main electronics temperature" submenu
Navigation

 Expert → Diagnostics → Min/max val. → Main elect.temp.

‣ Main electronics temperature
Minimum value→  175

Maximum value→  175

Minimum value
Navigation

 Expert → Diagnostics → Min/max val. → Main elect.temp. → Minimum value

Description

Shows the lowest temperature measured to date for the main electronic module.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Temperature unit parameter (→  56)

Maximum value
Navigation

 Expert → Diagnostics → Min/max val. → Main elect.temp. → Maximum value

Description

Shows the highest temperature measured to date for the main electronic module.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Temperature unit parameter (→  56)

Endress+Hauser

175

Description of device parameters

Proline Teqwave M 300 HART

"Sensor electronics temperature (ISEM)" submenu
Navigation

 Expert → Diagnostics → Min/max val. → Sensor elec.temp‣ Sensor electronics temperature(ISEM)

Minimum value→  176

Maximum value→  176

Minimum value
Navigation

 Expert → Diagnostics → Min/max val. → Sensor elec.temp → Minimum value

Description

Shows the lowest temperature measured to date for the sensor electronic module.

User interface

Signed floating-point number

Maximum value
Navigation

 Expert → Diagnostics → Min/max val. → Sensor elec.temp → Maximum value

Description

Shows the highest temperature measured to date for the sensor electronic module.

User interface

Signed floating-point number"Medium temperature" submenu
Navigation

 Expert → Diagnostics → Min/max val. → Medium temp.

‣ Medium temperature

176

Minimum value→  177

Maximum value→  177

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Minimum value
Navigation

 Expert → Diagnostics → Min/max val. → Medium temp. → Minimum value

Description

Shows the lowest medium temperature measured to date.

User interface

Positive floating-point number

Maximum value
Navigation

 Expert → Diagnostics → Min/max val. → Medium temp. → Maximum value

Description

Shows the highest medium temperature measured to date.

User interface

Positive floating-point number"Total solids" submenu
Navigation

 Expert → Diagnostics → Min/max val. → Total solids‣ Total solids
Maximum value→  177

Minimum value→  177

Maximum value
Navigation

 Expert → Diagnostics → Min/max val. → Total solids → Maximum value

Description

Shows the highest total solids value measured to date.

User interface

Positive floating-point number

Minimum value
Navigation

 Expert → Diagnostics → Min/max val. → Total solids → Minimum value

Description

Shows the lowest total solids value measured to date.

Endress+Hauser

177

Description of device parameters

User interface

Proline Teqwave M 300 HART

Positive floating-point number"Conductivity" submenu
Navigation

 Expert → Diagnostics → Min/max val. → Conductivity‣ Conductivity
Maximum value→  178

Minimum value→  178

Maximum value
Navigation

 Expert → Diagnostics → Min/max val. → Conductivity → Maximum value

Description

Shows the highest conductivity measured to date.

User interface

Positive floating-point number

Minimum value
Navigation

 Expert → Diagnostics → Min/max val. → Conductivity → Minimum value

Description

Shows the lowest conductivity measured to date.

User interface

Positive floating-point number

3.8.11"Heartbeat Technology" submenu

For detailed information on the parameter descriptions of the Heartbeat Verification
application package, see the Special Documentation for the device →  7
Navigation

178

 Expert → Diagnostics → Heartbeat Techn.

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters3.8.12"Simulation" submenu

Navigation

 Expert → Diagnostics → Simulation‣ Simulation

Endress+Hauser

Assign simulation process variable→  180

Process variable value→  180

Current input 1 to n simulation→  180

Value current input 1 to n

→  181

Status input 1 to n simulation→  181

Input signal level 1 to n

→  182

Current output 1 to n simulation→  182

Current output value→  182

Frequency output 1 to n simulation→  183

Frequency output 1 to n value→  183

Pulse output simulation 1 to n

→  184

Pulse value 1 to n

→  184

Switch output simulation 1 to n

→  184

Switch state 1 to n

→  185

Relay output 1 to n simulation→  185

Switch state 1 to n

→  186

Device alarm simulation→  186

Diagnostic event category→  187

Diagnostic event simulation→  187179

Description of device parameters

Proline Teqwave M 300 HART

Assign simulation process variable

Navigation

 Expert → Diagnostics → Simulation → Assign proc.var.

Prerequisite

The Load rate option is only available if the volume flow of the medium is read in via theCurrent input 1 to n (→  49) or the fieldbus.

Description

Select a process variable for the simulation process that is activated.

Selection• Off• Load rate *• Total solids• Temperature• Electronics temperature• Conductivity• Corrected conductivity

Factory setting

Off

Process variable value

Navigation

 Expert → Diagnostics → Simulation → Proc. var. value

Description

Enter the simulation value for the selected process variable.

User entry

Signed floating-point number

Factory setting0

Current input 1 to n simulation



Navigation

 Expert → Diagnostics → Simulation → Curr.inp 1 to n sim.

Description

Option for switching simulation of the current input on and off. The display alternates
between the measured value and a diagnostic message of the "Function check" category (C)while simulation is in progress.
The desired simulation value is defined in the Value current input 1 to n parameter.

Selection

• Off
• On

Factory setting

Off

*

180

Visibility depends on order options or device settings

Endress+Hauser

Proline Teqwave M 300 HART

Additional information

Description of device parameters

Selection• Off
Current simulation is switched off. The device is in normal measuring mode or anotherprocess variable is being simulated.
• On
Current simulation is active.

Value current input 1 to n

Navigation

 Expert → Diagnostics → Simulation → Value curr.inp 1 to n

Prerequisite

In the Current input 1 to n simulation parameter, the On option is selected.

Description

Use this function to enter the current value for the simulation. In this way, users can verify
the correct configuration of the current input and the correct function of upstream feed-inunits.

User entry

0 to 22.5 mA

Status input 1 to n simulation



Navigation

 Expert → Diagnostics → Simulation → Status inp 1 to n sim

Description

Use this function to switch simulation of the status input on and off. The display alternates
between the measured value and a diagnostic message of the "Function check" category (C)while simulation is in progress.

Selection

• Off
• On

Factory setting

Off

Additional information

Description
The desired simulation value is defined in the Input signal level parameter(→  182).
Selection• Off
Simulation for the status input is switched off. The device is in normal measuring modeor another process variable is being simulated.
• On
Simulation for the status input is active.

Endress+Hauser

181

Description of device parameters

Proline Teqwave M 300 HART

Input signal level 1 to n

Navigation

 Expert → Diagnostics → Simulation → Signal level 1 to n

Prerequisite

In the Status input simulation parameter (→  181), the On option is selected.

Description

Use this function to select the signal level for the simulation of the status input. In thisway, users can verify the correct configuration of the status input and the correct functionof upstream feed-in units.

Selection• High• Low

Current output 1 to n simulation



Navigation

 Expert → Diagnostics → Simulation → Curr.outp 1 to n sim.

Description

Use this function to switch simulation of the current output on and off. The display
alternates between the measured value and a diagnostic message of the "Function check"
category (C) while simulation is in progress.

Selection

• Off
• On

Factory setting

Off

Additional information

Description
The desired simulation value is defined in the Value current output 1 to n parameter.
Selection• Off
Current simulation is switched off. The device is in normal measuring mode or anotherprocess variable is being simulated.
• On
Current simulation is active.

Current output value

Navigation

 Expert → Diagnostics → Simulation → Curr.outp val.

Prerequisite

In the Current output 1 to n simulation parameter, the On option is selected.

Description

Use this function to enter a current value for the simulation. In this way, users can verify
the correct adjustment of the current output and the correct function of downstream
switching units.

User entry

3.59 to 22.5 mA

182

Endress+Hauser

Proline Teqwave M 300 HART

Additional information

Description of device parameters

Dependency
The input range is dependent on the option selected in the Current span parameter(→  75).

Frequency output 1 to n simulation



Navigation

 Expert → Diagnostics → Simulation → Freq.outp 1 to n sim.

Prerequisite

In the Operating mode parameter (→  87), the Frequency option is selected.

Description

Use this function to switch simulation of the frequency output on and off. The display
alternates between the measured value and a diagnostic message of the "Function check"
category (C) while simulation is in progress.

Selection

• Off
• On

Factory setting

Off

Additional information

Description
The desired simulation value is defined in the Frequency value 1 to n parameter.
Selection• Off
Frequency simulation is switched off. The device is in normal measuring mode oranother process variable is being simulated.
• On
Frequency simulation is active.

Frequency output 1 to n value

Navigation

 Expert → Diagnostics → Simulation → Freq.outp 1 to n val.

Prerequisite

In the Frequency simulation 1 to n parameter, the On option is selected.

Description

Use this function to enter a frequency value for the simulation. In this way, users can verify
the correct adjustment of the frequency output and the correct function of downstream
switching units.

User entry

0.0 to 12 500.0 Hz

Endress+Hauser

183

Description of device parameters

Proline Teqwave M 300 HART

Pulse output simulation 1 to n

Navigation

 Expert → Diagnostics → Simulation → Puls.outp.sim. 1 to n

Prerequisite

In the Operating mode parameter (→  87), the Pulse option is selected.

Description

Use this function to switch simulation of the pulse output on and off. The display
alternates between the measured value and a diagnostic message of the "Function check"
category (C) while simulation is in progress.

Selection• Off• Fixed value• Down-counting value

Factory setting

Off

Additional information

Description
The desired simulation value is defined in the Pulse value 1 to n parameter.
Selection• Off
Pulse simulation is switched off. The device is in normal measuring mode or anotherprocess variable is being simulated.
• Fixed value
Pulses are continuously output with the pulse width specified in the Pulse widthparameter (→  89).
• Down-counting value
The pulses specified in the Pulse value parameter (→  184) are output.

Pulse value 1 to n

Navigation

 Expert → Diagnostics → Simulation → Pulse value 1 to n

Prerequisite

In the Pulse output simulation 1 to n parameter, the Down-counting value option is
selected.

Description

Use this function to enter a pulse value for the simulation. In this way, users can verify thecorrect adjustment of the pulse output and the correct function of downstream switchingunits.

User entry

0 to 65 535

Switch output simulation 1 to n

Navigation

 Expert → Diagnostics → Simulation → Switch sim. 1 to n

Prerequisite

In the Operating mode parameter (→  87), the Switch option is selected.

184

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Description

Use this function to switch simulation of the switch output on and off. The display
alternates between the measured value and a diagnostic message of the "Function check"
category (C) while simulation is in progress.

Selection

• Off
• On

Factory setting

Off

Additional information

Description
The desired simulation value is defined in the Switch state 1 to n parameter.
Selection• Off
Switch simulation is switched off. The device is in normal measuring mode or anotherprocess variable is being simulated.
• On
Switch simulation is active.

Switch state 1 to n

Navigation

 Expert → Diagnostics → Simulation → Switch state 1 to n

Description

Use this function to select a switch value for the simulation. In this way, users can verify
the correct adjustment of the switch output and the correct function of downstream
switching units.

Selection• Open• Closed

Additional information

Selection• OpenSwitch simulation is switched off. The device is in normal measuring mode or anotherprocess variable is being simulated.
• Closed
Switch simulation is active.

Relay output 1 to n simulation



Navigation

 Expert → Diagnostics → Simulation → Relay out. 1 to n sim

Description

Use this function to switch simulation of the relay output on and off. The display
alternates between the measured value and a diagnostic message of the "Function check"
category (C) while simulation is in progress.

Selection

• Off
• On

Factory setting

Off

Endress+Hauser

185

Description of device parameters

Additional information

Proline Teqwave M 300 HART

Description
The desired simulation value is defined in the Switch state 1 to n parameter.
Selection• Off
Relay simulation is switched off. The device is in normal measuring mode or anotherprocess variable is being simulated.
• On
Relay simulation is active.

Switch state 1 to n

Navigation

 Expert → Diagnostics → Simulation → Switch state 1 to n

Prerequisite

The On option is selected in the Switch output simulation 1 to n parameter parameter.

Description

Use this function to select a relay value for the simulation. In this way, users can verify thecorrect adjustment of the relay output and the correct function of downstream switchingunits.

Selection• Open• Closed

Additional information

Selection• OpenRelay simulation is switched off. The device is in normal measuring mode or anotherprocess variable is being simulated.
• Closed
Relay simulation is active.

Device alarm simulation



Navigation

 Expert → Diagnostics → Simulation → Dev. alarm sim.

Description

Use this function to switch the device alarm on and off.

Selection

• Off
• On

Factory setting

Off

Additional information

Description
The display alternates between the measured value and a diagnostic message of the"Function check" category (C) while simulation is in progress.

186

Endress+Hauser

Proline Teqwave M 300 HART

Description of device parameters

Diagnostic event category



Navigation

Expert → Diagnostics → Simulation → Event category

Description

Use this function to select the category of the diagnostic events that are displayed for thesimulation in the Diagnostic event simulation parameter (→  187).

Selection• Sensor• Electronics• Configuration• ProcessFactory setting

Process

Diagnostic event simulation



Navigation

 Expert → Diagnostics → Simulation → Diagnostic event

Description

Use this function to select a diagnostic event for the simulation process that is activated.

Selection• Off• Diagnostic event picklist (depends on the category selected)

Factory setting

Off

Additional information

Description
For the simulation, you can choose from the diagnostic events of the category selectedin the Diagnostic event category parameter (→  187).

Endress+Hauser

187

Country-specific factory settings

Proline Teqwave M 300 HART

4Country-specific factory settings

4.1

SI units
The country-specific factory setting in SI units is made for all countries except the USA
and Canada.

4.1.1

System units

Process variable

Unit

Total solids%TS

Densityg/l

Mass flow

kg/h

Mass

kg

Volumel/hTemperature°C

ConductivityµS/cm

Load rate

kg/h

For further information on system units, see: System units submenu (→  54)

4.1.2

Output current span

OutputCurrent range

Current output 1 to n

4 to 20 mA NAMUR

For further information on the current ranges, see: Current range output parameter(→  75)

4.2

US units
The country-specific factory setting in US units is made for the USA and Canada.

4.2.1

188

System units

Process variable

Unit

Total solids%TS

Densitylb/ft³Mass flow

lb/h

Mass

lb

Volumel/hTemperature°F

Endress+Hauser

Proline Teqwave M 300 HARTCountry-specific factory settings

Process variable

Unit

ConductivityµS/cm

Load ratelb/h

For further information on system units, see: System units submenu (→  54)

4.2.2

Output current span

OutputCurrent range

Current output 1 to n

4 to 20 mA US

For further information on the current ranges, see: Current range output parameter(→  75)

Endress+Hauser

189

Index

Proline Teqwave M 300 HART

Index
0…9

0/4 mA value (Parameter) . . . . . . . . . . . . . . . . . . . . . 70
0% bargraph value 1 (Parameter) . . . . . . . . . . . . . . . 17
0% bargraph value 3 (Parameter) . . . . . . . . . . . . . . . 20
0% bargraph value 5 (Parameter) . . . . . . . . . . . . . . . 24
0% bargraph value 7 (Parameter) . . . . . . . . . . . . . . . 26
2.4 GHz WLAN channel (Parameter) . . . . . . . . . . . . 144
20 mA value (Parameter) . . . . . . . . . . . . . . . . . . . . . 70
100% bargraph value 1 (Parameter) . . . . . . . . . . . . . 18
100% bargraph value 3 (Parameter) . . . . . . . . . . . . . 21
100% bargraph value 5 (Parameter) . . . . . . . . . . . . . 24
100% bargraph value 7 (Parameter) . . . . . . . . . . . . . 27

A

Activate SW option (Parameter) . . . . . . . . . . . . . . . . 44
Active level (Parameter) . . . . . . . . . . . . . . . . . . . . . . 73
Actual diagnostics (Parameter) . . . . . . . . . . . . . . . . 151
Administration (Submenu) . . . . . . . . . . . . . . . . . . . . 41
Alarm delay (Parameter) . . . . . . . . . . . . . . . . . . . . . . 35
Application (Submenu) . . . . . . . . . . . . . . . . . . . . . . 146
Apply I/O configuration (Parameter) . . . . . . . . . . . . . 67
Assign behavior of diagnostic no. 302 (Parameter) . . . 38
Assign behavior of diagnostic no. 441 (Parameter) . . . 37
Assign behavior of diagnostic no. 442 (Parameter) . . . 37
Assign behavior of diagnostic no. 443 (Parameter) . . . 37
Assign behavior of diagnostic no. 444 (Parameter) . . . 38
Assign behavior of diagnostic no. 832 (Parameter) . . . 38
Assign behavior of diagnostic no. 833 (Parameter) . . . 39
Assign behavior of diagnostic no. 834 (Parameter) . . . 39
Assign behavior of diagnostic no. 835 (Parameter) . . . 40
Assign behavior of diagnostic no. 907 (Parameter) . . . 40
Assign behavior of diagnostic no. 908 (Parameter) . . . 40
Assign channel 1 (Parameter) . . . . . . . . . . . . . . . . . 167
Assign channel 2 (Parameter) . . . . . . . . . . . . . . . . . 168
Assign channel 3 (Parameter) . . . . . . . . . . . . . . . . . 168
Assign channel 4 (Parameter) . . . . . . . . . . . . . . . . . 168
Assign diagnostic behavior (Parameter) . . . . . . . 97, 106
Assign flow direction check (Parameter) . . . . . . 101, 105
Assign frequency output (Parameter) . . . . . . . . . . . . . 92
Assign limit (Parameter) . . . . . . . . . . . . . . . . . . 98, 105
Assign process variable (Parameter) . . . . . . . . . . . . . 59
Assign process variable 1 (Parameter) . . . . . . . . . . . 147
Assign pulse output (Parameter) . . . . . . . . . . . . . . . . 88
Assign PV (Parameter) . . . . . . . . . . . . . . . . . . . . . . 128
Assign QV (Parameter) . . . . . . . . . . . . . . . . . . . . . . 130
Assign simulation process variable (Parameter) . . . . 180
Assign SSID name (Parameter) . . . . . . . . . . . . . . . . 143
Assign status (Parameter) . . . . . . . . . . . . . . . . 101, 106
Assign status input (Parameter) . . . . . . . . . . . . . . . . . 72
Assign SV (Parameter) . . . . . . . . . . . . . . . . . . . . . . 128
Assign TV (Parameter) . . . . . . . . . . . . . . . . . . . . . . 129

B

Backlight (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 32
Backup state (Parameter) . . . . . . . . . . . . . . . . . . . . . 34

190

Bootloader revision (Parameter). . . . . . . . . . . . . . . . . . . . . . . . 162, 163, 164, 165, 166
Build no. software (Parameter) 161, 162, 164, 165, 166
Burst command (Parameter) . . . . . . . . . . . . . . . . . . 112
Burst command 1 to n (Parameter) . . . . . . . . . . . . . 119
Burst configuration 1 to n (Submenu) . . . . . . . . . . . 117
Burst mode 1 to n (Parameter) . . . . . . . . . . . . . . . . 118
Burst trigger level (Parameter) . . . . . . . . . . . . . . . . . 123
Burst trigger mode (Parameter) . . . . . . . . . . . . . . . . 122
Burst variable 0 (Parameter) . . . . . . . . . . . . . . . . . . 119
Burst variable 1 (Parameter) . . . . . . . . . . . . . . . . . . 120
Burst variable 2 (Parameter) . . . . . . . . . . . . . . . . . . 120
Burst variable 3 (Parameter) . . . . . . . . . . . . . . . . . . 121
Burst variable 4 (Parameter) . . . . . . . . . . . . . . . . . . 121
Burst variable 5 (Parameter) . . . . . . . . . . . . . . . . . . 121
Burst variable 6 (Parameter) . . . . . . . . . . . . . . . . . . 121
Burst variable 7 (Parameter) . . . . . . . . . . . . . . . . . . 122

C

Capture mode (Parameter) . . . . . . . . . . . . . . . . . . . 110
Clear logging data (Parameter) . . . . . . . . . . . . . . . . 169
Communication (Submenu) . . . . . . . . . . . . . . . . . . . 109
Comparison result (Parameter) . . . . . . . . . . . . . . . . . 34
Conductivity (Parameter) . . . . . . . . . . . . . . . . . . . . . . 47
Conductivity (Submenu) . . . . . . . . . . . . . . . . . . . . . 178
Conductivity damping (Parameter) . . . . . . . . . . . . . . . 58
Conductivity factor (Parameter) . . . . . . . . . . . . . . . . . 63
Conductivity offset (Parameter) . . . . . . . . . . . . . . . . . 63
Conductivity unit (Parameter) . . . . . . . . . . . . . . . . . . 57
Configuration (Submenu) . . . . . . . . . . . . . . . . 110, 115
Configuration backup (Submenu) . . . . . . . . . . . . . . . . 32
Configuration counter (Parameter) . . . . . . . . . . . . . 160
Configuration management (Parameter) . . . . . . . . . . 33
Confirm access code (Parameter) . . . . . . . . . . . . . . . . 42
Connection state (Parameter) . . . . . . . . . . . . . . . . . 144
Contrast display (Parameter) . . . . . . . . . . . . . . . . . . . 31
Corrected conductivity (Parameter) . . . . . . . . . . . . . . 47
Corrected conductivity factor (Parameter) . . . . . . . . . 64
Corrected conductivity offset (Parameter) . . . . . . . . . . 64
Current input 1 to n (Submenu) . . . . . . . . . . . . . . 49, 68
Current input 1 to n simulation (Parameter) . . . . . . . 180
Current output 1 to n (Submenu) . . . . . . . . . . . . . . . . 74
Current output 1 to n simulation (Parameter) . . . . . . 182
Current output value (Parameter) . . . . . . . . . . . . . . 182
Current range output (Parameter) . . . . . . . . . . . . . . . 75
Current span (Parameter) . . . . . . . . . . . . . . . . . . . . . 69

D

Damping current output (Parameter) . . . . . . . . . . . . . 83
Damping output (Parameter) . . . . . . . . . . . . . . . . . . . 94
Data logging (Parameter) . . . . . . . . . . . . . . . . . . . . 170
Data logging (Submenu) . . . . . . . . . . . . . . . . . . . . . 166
Data logging control (Parameter) . . . . . . . . . . . . . . . 170
Data logging status (Parameter) . . . . . . . . . . . . . . . 171
Date/time (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 66
Date/time format (Parameter) . . . . . . . . . . . . . . . . . . 57
Endress+Hauser

Proline Teqwave M 300 HART

Decimal places 1 (Parameter) . . . . . . . . . . . . . . . . . . 18
Decimal places 2 (Parameter) . . . . . . . . . . . . . . . . . . 19
Decimal places 3 (Parameter) . . . . . . . . . . . . . . . . . . 21
Decimal places 4 (Parameter) . . . . . . . . . . . . . . . . . . 22
Decimal places 5 (Parameter) . . . . . . . . . . . . . . . . . . 24
Decimal places 6 (Parameter) . . . . . . . . . . . . . . . . . . 25
Decimal places 7 (Parameter) . . . . . . . . . . . . . . . . . . 27
Decimal places 8 (Parameter) . . . . . . . . . . . . . . . . . . 28
Default gateway (Parameter) . . . . . . . . . . . . . . . . . . 138
Define access code (Parameter) . . . . . . . . . . . . . . . . . 41
Define access code (Wizard) . . . . . . . . . . . . . . . . . . . 41
Density unit (Parameter) . . . . . . . . . . . . . . . . . . . . . . 55
Device alarm simulation (Parameter) . . . . . . . . . . . . 186
Device ID (Parameter) . . . . . . . . . . . . . . . . . . . 111, 125
Device information (Submenu) . . . . . . . . . . . . . . . . 157
Device name (Parameter) . . . . . . . . . . . . . . . . . . . . 159
Device reset (Parameter) . . . . . . . . . . . . . . . . . . . . . . 43
Device revision (Parameter) . . . . . . . . . . . . . . . . . . . 124
Device tag (Parameter) . . . . . . . . . . . . . . . . . . 116, 158
Device type (Parameter) . . . . . . . . . . . . . . . . . 111, 125
DHCP client (Parameter) . . . . . . . . . . . . . . . . . . . . . 137
Diagnostic behavior (Submenu) . . . . . . . . . . . . . . . . . 36
Diagnostic configuration (Submenu) . . . . . . . . . . . . 131
Diagnostic event category (Parameter) . . . . . . . . . . . 187
Diagnostic event simulation (Parameter) . . . . . . . . . 187
Diagnostic handling (Submenu) . . . . . . . . . . . . . . . . . 35
Diagnostic list (Submenu) . . . . . . . . . . . . . . . . . . . . 153
Diagnostics (Submenu) . . . . . . . . . . . . . . . . . . . . . . 150
Diagnostics 1 (Parameter) . . . . . . . . . . . . . . . . . . . . 153
Diagnostics 2 (Parameter) . . . . . . . . . . . . . . . . . . . . 154
Diagnostics 3 (Parameter) . . . . . . . . . . . . . . . . . . . . 154
Diagnostics 4 (Parameter) . . . . . . . . . . . . . . . . . . . . 154
Diagnostics 5 (Parameter) . . . . . . . . . . . . . . . . . . . . 155
Direct access
0% bargraph value 5 . . . . . . . . . . . . . . . . . . . . . . . 24
0% bargraph value 7 . . . . . . . . . . . . . . . . . . . . . . . 26
2.4 GHz WLAN channel . . . . . . . . . . . . . . . . . . . 144
100% bargraph value 5 . . . . . . . . . . . . . . . . . . . . 24
100% bargraph value 7 . . . . . . . . . . . . . . . . . . . . 27
Assign SSID name . . . . . . . . . . . . . . . . . . . . . . . 143
Connection state . . . . . . . . . . . . . . . . . . . . . . . . 144
Decimal places 5 . . . . . . . . . . . . . . . . . . . . . . . . . . 24
Decimal places 6 . . . . . . . . . . . . . . . . . . . . . . . . . . 25
Decimal places 7 . . . . . . . . . . . . . . . . . . . . . . . . . . 27
Decimal places 8 . . . . . . . . . . . . . . . . . . . . . . . . . . 28
Gateway IP address . . . . . . . . . . . . . . . . . . . . . . 145
IP address domain name server . . . . . . . . . . . . . . 145
Network security . . . . . . . . . . . . . . . . . . . . . . . . 141
Received signal strength . . . . . . . . . . . . . . . . . . . 145
Relay output 1 to n simulation . . . . . . . . . . . . . . 185
Security identification . . . . . . . . . . . . . . . . . . . . . 141
Select antenna . . . . . . . . . . . . . . . . . . . . . . . . . . 144
SSID name . . . . . . . . . . . . . . . . . . . . . . . . . 140, 144
Switch state 1 to n . . . . . . . . . . . . . . . . . . . . . . . 186
Timestamp . . . . . . . . . . . . . . . . . . . . . . . . . 151, 152
User name . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 141
Value 5 display . . . . . . . . . . . . . . . . . . . . . . . . . . . 23
Value 6 display . . . . . . . . . . . . . . . . . . . . . . . . . . . 25
Value 7 display . . . . . . . . . . . . . . . . . . . . . . . . . . . 26
Endress+Hauser

Index

Value 8 display . . . . . . . . . . . . . . . . . . . . . . . . . . . 28
WLAN . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 140
WLAN IP address . . . . . . . . . . . . . . . . . . . . . . . . 142
WLAN MAC address . . . . . . . . . . . . . . . . . . . . . 142
WLAN mode . . . . . . . . . . . . . . . . . . . . . . . . . . . 140
WLAN passphrase . . . . . . . . . . . . . . . . . . . . . . . 143
WLAN password . . . . . . . . . . . . . . . . . . . . . . . . 142
WLAN subnet mask . . . . . . . . . . . . . . . . . . . . . . 143
Display (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . 13
Display channel 1 (Submenu) . . . . . . . . . . . . . . . . . 172
Display channel 2 (Submenu) . . . . . . . . . . . . . . . . . 172
Display channel 3 (Submenu) . . . . . . . . . . . . . . . . . 173
Display channel 4 (Submenu) . . . . . . . . . . . . . . . . . 173
Display damping (Parameter) . . . . . . . . . . . . . . . . . . 29
Display interval (Parameter) . . . . . . . . . . . . . . . . . . . 29
Display language (Parameter) . . . . . . . . . . . . . . . . . . 23
Display module (Submenu) . . . . . . . . . . . . . . . . . . . 165
Document
Design . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Explanation of the structure of a parameter
description . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6
Function . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Symbols used . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6
Target group . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Using the document . . . . . . . . . . . . . . . . . . . . . . . . 4
Document function . . . . . . . . . . . . . . . . . . . . . . . . . . . 4

EElectronics temperature (Parameter) . . . . . . . . . . . . . 47
Electronics temperature (Submenu) . . . . . . . . . . . . . 174
Electronics temperature factor (Parameter) . . . . . . . . 64
Electronics temperature offset (Parameter) . . . . . . . . 64
ENP version (Parameter) . . . . . . . . . . . . . . . . . . . . . 160
Enter access code (Parameter) . . . . . . . . . . . . . . . . . . 13
Entire logging duration (Parameter) . . . . . . . . . . . . 171
Event category 441 (Parameter) . . . . . . . . . . . . . . . 132
Event category 442 (Parameter) . . . . . . . . . . . . . . . 132
Event category 443 (Parameter) . . . . . . . . . . . . . . . 133
Event category 444 (Parameter) . . . . . . . . . . . . . . . 133
Event category 832 (Parameter) . . . . . . . . . . . . . . . 134
Event category 833 (Parameter) . . . . . . . . . . . . . . . 134
Event category 834 (Parameter) . . . . . . . . . . . . . . . 134
Event category 835 (Parameter) . . . . . . . . . . . . . . . 135
Event list (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 156
Event logbook (Submenu) . . . . . . . . . . . . . . . . . . . . 155
Expert (Menu) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 11
Extended order code 1 (Parameter) . . . . . . . . . . . . . 159
Extended order code 2 (Parameter) . . . . . . . . . . . . . 160
Extended order code 3 (Parameter) . . . . . . . . . . . . . 160
External compensation (Submenu) . . . . . . . . . . . . . . . 61

F

Factory adjustment (Submenu) . . . . . . . . . . . . . . . . . 65
Factory settings . . . . . . . . . . . . . . . . . . . . . . . . . . . . 188
SI units . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 188
US units . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 188
Failure behavior current output (Parameter) . . . . . . . 83
Failure current (Parameter) . . . . . . . . . . . . . . . . . . . . 84
Failure frequency (Parameter) . . . . . . . . . . . . . . . . . . 96
191

Index

Failure mode (Parameter) . . . . 71, 91, 95, 102, 108, 113
Failure value (Parameter) . . . . . . . . . . . . . . . . . 71, 114
Fieldbus writing access (Parameter) . . . . . . . . . . . . . 117
Filter options (Parameter) . . . . . . . . . . . . . . . . . . . . 156
Firmware version (Parameter). . . . . . . . . . . . . . . . . . . . 159, 161, 162, 163, 165, 166
Fixed current (Parameter) . . . . . . . . . . . . . . . . . . . . . 76
Format display (Parameter) . . . . . . . . . . . . . . . . . . . . 14
Frequency output 1 to n simulation (Parameter) . . . . 183
Frequency output 1 to n value (Parameter) . . . . . . . . 183
Function
see Parameter

G

Gateway IP address (Parameter) . . . . . . . . . . . . . . . 145

H

Hardware revision (Parameter) . . . . . . . . . . . . . . . . 126
HART address (Parameter) . . . . . . . . . . . . . . . . . . . 116
HART date code (Parameter) . . . . . . . . . . . . . . . . . . 127
HART descriptor (Parameter) . . . . . . . . . . . . . . . . . . 126
HART input (Submenu) . . . . . . . . . . . . . . . . . . . . . . 110
HART message (Parameter) . . . . . . . . . . . . . . . . . . . 126
HART output (Submenu) . . . . . . . . . . . . . . . . . . . . . 115
HART revision (Parameter) . . . . . . . . . . . . . . . . . . . 125
HART short tag (Parameter) . . . . . . . . . . . . . . . . . . 116
Header (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . . . 30
Header text (Parameter) . . . . . . . . . . . . . . . . . . . . . . 30
Heartbeat Technology (Submenu) . . . . . . . . . . . . . . 178

I

I/O alteration code (Parameter) . . . . . . . . . . . . . . . . . 68
I/O configuration (Submenu) . . . . . . . . . . . . . . . . . . . 66
I/O module 1 to n information (Parameter) . . . . . . . . 66
I/O module 1 to n terminal numbers (Parameter) . . . . 66
I/O module 1 to n type (Parameter) . . . . . . . . . . . . . . 67
I/O module 2 (Submenu) . . . . . . . . . . . . . . . . . . . . . 163
I/O module 2 terminal numbers (Parameter) . . 163, 164
I/O module 3 (Submenu) . . . . . . . . . . . . . . . . . . . . . 164
I/O module 3 terminal numbers (Parameter) . . 163, 164
I/O module 4 terminal numbers (Parameter) . . 163, 164
Information (Submenu) . . . . . . . . . . . . . . . . . . . . . . 124
Input (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 68, 114
Input signal level 1 to n (Parameter) . . . . . . . . . . . . 182
Input values (Submenu) . . . . . . . . . . . . . . . . . . . . . . . 49
Invert output signal (Parameter) . . . . . . . . . . . . . . . 103
IP address (Parameter) . . . . . . . . . . . . . . . . . . . . . . 137
IP address domain name server (Parameter) . . . . . . . 145

L

Last backup (Parameter) . . . . . . . . . . . . . . . . . . . . . . 33
Load rate (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 47
Load rate factor (Parameter) . . . . . . . . . . . . . . . . . . . 65
Load rate offset (Parameter) . . . . . . . . . . . . . . . . . . . 65
Locking status (Parameter) . . . . . . . . . . . . . . . . . . . . 11
Logging delay (Parameter) . . . . . . . . . . . . . . . . . . . 170
Logging interval (Parameter) . . . . . . . . . . . . . . . . . . 169
Login page (Parameter) . . . . . . . . . . . . . . . . . . . . . . 138
Lower range limit (Parameter) . . . . . . . . . . . . . . . . . . 59
Lower range value output (Parameter) . . . . . . . . . . . . 77
192

Proline Teqwave M 300 HARTM

MAC address (Parameter) . . . . . . . . . . . . . . . . . . . . 136
Main electronic module + I/O module 1 (Submenu) . 161
Main electronics temperature (Submenu) . . . . . . . . . 175
Manufacturer ID (Parameter) . . . . . . . . . . . . . . 112, 125
Mass flow unit (Parameter) . . . . . . . . . . . . . . . . . . . . 55
Mass unit (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 56
Max. switch cycles number (Parameter) . . . . . . . . . . . 54
Max. update period (Parameter) . . . . . . . . . . . . . . . 123
Maximum frequency value (Parameter) . . . . . . . . . . . 93
Maximum value (Parameter) . . . . . . 175, 176, 177, 178
Measured current (Parameter) . . . . . . . . . . . . . . . 51, 85
Measured current 1 to n (Parameter) . . . . . . . . . . . . . 50
Measured values (Submenu) . . . . . . . . . . . . . . . . . . . 46
Measured values 1 to n (Parameter) . . . . . . . . . . . . . . 49
Measuring interval (Parameter) . . . . . . . . . . . . . . . . . 62
Measuring mode (Parameter) . . . . . . . . . . . . . . . 90, 94
Measuring mode current output (Parameter) . . . . . . . 78
Measuring value at maximum frequency (Parameter) 93
Measuring value at minimum frequency (Parameter) 93
Medium temperature (Submenu) . . . . . . . . . . . . . . . 176
Menu
Expert . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 11
Min. update period (Parameter) . . . . . . . . . . . . . . . . 123
Min/max values (Submenu) . . . . . . . . . . . . . . . . . . . 174
Minimum frequency value (Parameter) . . . . . . . . . . . 92
Minimum value (Parameter) . . 174, 175, 176, 177, 178

N

Network security (Parameter) . . . . . . . . . . . . . . . . . 141
No. of preambles (Parameter) . . . . . . . . . . . . . . . . . 117
Nominal diameter (Parameter) . . . . . . . . . . . . . . . . . 65

O

Operating mode (Parameter) . . . . . . . . . . . . . . . . . . . 87
Operating time (Parameter) . . . . . . . . . . . . . 32, 42, 153
Operating time from restart (Parameter) . . . . . . . . . 152
Order code (Parameter) . . . . . . . . . . . . . . . . . . . . . . 159
Output (Submenu) . . . . . . . . . . . . . . . . . . . . . . . 73, 127
Output current (Parameter) . . . . . . . . . . . . . . . . . 51, 84
Output frequency (Parameter) . . . . . . . . . . . . . . . 52, 96
Output values (Submenu) . . . . . . . . . . . . . . . . . . . . . 50

PParameter
Structure of description . . . . . . . . . . . . . . . . . . . . . 6
Partially filled pipe detection (Parameter) . . . . . . . . . 60
Partially filled pipe detection (Submenu) . . . . . . . . . . 60
Powerless relay status (Parameter) . . . . . . . . . . . . . 109
Preset value 1 (Parameter) . . . . . . . . . . . . . . . . . . . 149
Previous diagnostics (Parameter) . . . . . . . . . . . . . . . 152
Primary variable (PV) (Parameter) . . . . . . . . . . . . . . 128
Process parameters (Submenu) . . . . . . . . . . . . . . . . . 57
Process variable adjustment (Submenu) . . . . . . . . . . . 62
Process variable current output (Parameter) . . . . . . . . 75
Process variable unit 1 (Parameter) . . . . . . . . . . . . . 147
Process variable value (Parameter) . . . . . . . . . . . . . . 180
Process variables (Submenu) . . . . . . . . . . . . . . . . . . . 46
Pulse output (Parameter) . . . . . . . . . . . . . . . . . . . 52, 91
Endress+Hauser

Proline Teqwave M 300 HART

Pulse output simulation 1 to n (Parameter) . . . . . . . 184
Pulse scaling (Parameter) . . . . . . . . . . . . . . . . . . . . . 89
Pulse value 1 to n (Parameter) . . . . . . . . . . . . . . . . . 184
Pulse width (Parameter) . . . . . . . . . . . . . . . . . . . . . . 89
Pulse/frequency/switch output 1 to n (Submenu) 51, 85

Q

Quaternary variable (QV) (Parameter) . . . . . . . . . . . 131

R

Received signal strength (Parameter) . . . . . . . . . . . . 145
Relay output 1 to n (Submenu) . . . . . . . . . . . . . . 53, 103
Relay output 1 to n simulation (Parameter) . . . . . . . 185
Relay output function (Parameter) . . . . . . . . . . . . . . 104
Reset access code (Parameter) . . . . . . . . . . . . . . . . . . 43
Reset access code (Submenu) . . . . . . . . . . . . . . . . . . . 42
Reset all totalizers (Parameter) . . . . . . . . . . . . . . . . 146
Response time (Parameter) . . . . . . . . . . . . . . 60, 61, 95
Response time status input (Parameter) . . . . . . . . . . . 73

S

Secondary variable (SV) (Parameter) . . . . . . . . . . . . 129
Security identification (Parameter) . . . . . . . . . . . . . . 141
Select antenna (Parameter) . . . . . . . . . . . . . . . . . . . 144
Sensor (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . . 45
Sensor adjustment (Submenu) . . . . . . . . . . . . . . . . . . 62
Sensor electronic module (ISEM) (Submenu) . . . . . . 162
Sensor electronics temperature (ISEM) (Submenu) . . 176
Separator (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 31
Serial number (Parameter) . . . . . . . . . . . . . . . . . . . 158
Signal mode (Parameter) . . . . . . . . . . . . . . . . 69, 75, 87
Simulation (Submenu) . . . . . . . . . . . . . . . . . . . . . . . 179
Slot number (Parameter) . . . . . . . . . . . . . . . . . . . . . 113
Software option overview (Parameter) . . . . . . . . . . . . 45
Software revision (Parameter) . . . . . . . . . . . . . . . . . 126
SSID name (Parameter) . . . . . . . . . . . . . . . . . . 140, 144
Status (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . . 115
Status input 1 to n (Submenu) . . . . . . . . . . . . . . . . . . 71
Status input 1 to n simulation (Parameter) . . . . . . . . 181
Submenu
Administration . . . . . . . . . . . . . . . . . . . . . . . . . . . 41
Application . . . . . . . . . . . . . . . . . . . . . . . . . . . . 146
Burst configuration 1 to n . . . . . . . . . . . . . . . . . . 117
Communication . . . . . . . . . . . . . . . . . . . . . . . . . 109
Conductivity . . . . . . . . . . . . . . . . . . . . . . . . . . . . 178
Configuration . . . . . . . . . . . . . . . . . . . . . . . 110, 115
Configuration backup . . . . . . . . . . . . . . . . . . . . . . 32
Current input 1 to n . . . . . . . . . . . . . . . . . . . . 49, 68
Current output 1 to n . . . . . . . . . . . . . . . . . . . . . . 74
Data logging . . . . . . . . . . . . . . . . . . . . . . . . . . . 166
Device information . . . . . . . . . . . . . . . . . . . . . . . 157
Diagnostic behavior . . . . . . . . . . . . . . . . . . . . . . . 36
Diagnostic configuration . . . . . . . . . . . . . . . . . . . 131
Diagnostic handling . . . . . . . . . . . . . . . . . . . . . . . 35
Diagnostic list . . . . . . . . . . . . . . . . . . . . . . . . . . 153
Diagnostics . . . . . . . . . . . . . . . . . . . . . . . . . . . . 150
Display . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 13
Display channel 1 . . . . . . . . . . . . . . . . . . . . . . . . 172
Display channel 2 . . . . . . . . . . . . . . . . . . . . . . . . 172
Endress+Hauser

Index

Display channel 3 . . . . . . . . . . . . . . . . . . . . . . . . 173
Display channel 4 . . . . . . . . . . . . . . . . . . . . . . . . 173
Display module . . . . . . . . . . . . . . . . . . . . . . . . . . 165
Electronics temperature . . . . . . . . . . . . . . . . . . . 174
Event list . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 156
Event logbook . . . . . . . . . . . . . . . . . . . . . . . . . . 155
External compensation . . . . . . . . . . . . . . . . . . . . . 61
Factory adjustment . . . . . . . . . . . . . . . . . . . . . . . . 65
HART input . . . . . . . . . . . . . . . . . . . . . . . . . . . . 110
HART output . . . . . . . . . . . . . . . . . . . . . . . . . . . 115
Heartbeat Technology . . . . . . . . . . . . . . . . . . . . 178
I/O configuration . . . . . . . . . . . . . . . . . . . . . . . . . 66
I/O module 2 . . . . . . . . . . . . . . . . . . . . . . . . . . . 163
I/O module 3 . . . . . . . . . . . . . . . . . . . . . . . . . . . 164
Information . . . . . . . . . . . . . . . . . . . . . . . . . . . . 124
Input . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 68, 114
Input values . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 49
Main electronic module + I/O module 1 . . . . . . . . 161
Main electronics temperature . . . . . . . . . . . . . . . 175
Measured values . . . . . . . . . . . . . . . . . . . . . . . . . 46
Medium temperature . . . . . . . . . . . . . . . . . . . . . 176
Min/max values . . . . . . . . . . . . . . . . . . . . . . . . . 174
Output . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 73, 127
Output values . . . . . . . . . . . . . . . . . . . . . . . . . . . . 50
Partially filled pipe detection . . . . . . . . . . . . . . . . . 60
Process parameters . . . . . . . . . . . . . . . . . . . . . . . 57
Process variable adjustment . . . . . . . . . . . . . . . . . 62
Process variables . . . . . . . . . . . . . . . . . . . . . . . . . 46
Pulse/frequency/switch output 1 to n . . . . . . . 51, 85
Relay output 1 to n . . . . . . . . . . . . . . . . . . . . 53, 103
Reset access code . . . . . . . . . . . . . . . . . . . . . . . . . 42
Sensor . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 45
Sensor adjustment . . . . . . . . . . . . . . . . . . . . . . . . 62
Sensor electronic module (ISEM) . . . . . . . . . . . . . 162
Sensor electronics temperature (ISEM) . . . . . . . . 176
Simulation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 179
Status input 1 to n . . . . . . . . . . . . . . . . . . . . . . . . 71
System . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 13
System units . . . . . . . . . . . . . . . . . . . . . . . . . . . . 54
Total solids . . . . . . . . . . . . . . . . . . . . . . . . . . . . 177
Total solids monitoring . . . . . . . . . . . . . . . . . . . . . 58
Totalizer . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 48
Totalizer 1 to n . . . . . . . . . . . . . . . . . . . . . . . . . . 146
Value current output 1 to n . . . . . . . . . . . . . . . . . . 51
Value status input 1 to n . . . . . . . . . . . . . . . . . . . . 50
Web server . . . . . . . . . . . . . . . . . . . . . . . . . . . . 135
Subnet mask (Parameter) . . . . . . . . . . . . . . . . . . . . 137
Switch cycles (Parameter) . . . . . . . . . . . . . . . . . . . . . 54
Switch output function (Parameter) . . . . . . . . . . . . . . 96
Switch output simulation 1 to n (Parameter) . . . . . . 184
Switch state (Parameter) . . . . . . . . . . . . . . 53, 102, 109
Switch state 1 to n (Parameter) . . . . . . . . . . . . 185, 186
Switch-off delay (Parameter) . . . . . . . . . . . . . . 102, 107
Switch-off value (Parameter) . . . . . . . . . . . . . . 100, 107
Switch-on delay (Parameter) . . . . . . . . . . . . . . 101, 108
Switch-on value (Parameter) . . . . . . . . . . . . . . 100, 107
System (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . 13
System units (Submenu) . . . . . . . . . . . . . . . . . . . . . . 54

193

IndexT

Target group . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Temperature (Parameter) . . . . . . . . . . . . . . . . . . . . . 47
Temperature damping (Parameter) . . . . . . . . . . . . . . 58
Temperature factor (Parameter) . . . . . . . . . . . . . . . . 63
Temperature offset (Parameter) . . . . . . . . . . . . . . . . 63
Temperature unit (Parameter) . . . . . . . . . . . . . . . . . . 56
Terminal number (Parameter) . . . . . 69, 72, 74, 86, 104
Tertiary variable (TV) (Parameter) . . . . . . . . . . . . . . 130
Threshold (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 60
Timeout (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 113
Timestamp (Parameter) . . . . . . . . . . . . . . . . . . 151, 152
Total solids (Parameter) . . . . . . . . . . . . . . . . . . . . . . 46
Total solids (Submenu) . . . . . . . . . . . . . . . . . . . . . . 177
Total solids monitoring (Submenu) . . . . . . . . . . . . . . 58
Total solids override (Parameter) . . . . . . . . . . . . . . . . 58
Total solids unit (Parameter) . . . . . . . . . . . . . . . . . . . 55
Totalizer (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . 48
Totalizer 1 control (Parameter) . . . . . . . . . . . . . . . . 148
Totalizer 1 failure behavior (Parameter) . . . . . . . . . . 149
Totalizer 1 operation mode (Parameter) . . . . . . . . . . 148
Totalizer 1 overflow (Parameter) . . . . . . . . . . . . . . . . 48
Totalizer 1 to n (Submenu) . . . . . . . . . . . . . . . . . . . 146
Totalizer 1 value (Parameter) . . . . . . . . . . . . . . . . . . 48
Transmitter identifier (Parameter) . . . . . . . . . . . . . . . 44

Proline Teqwave M 300 HARTWLAN passphrase (Parameter) . . . . . . . . . . . . . . . . 143
WLAN password (Parameter) . . . . . . . . . . . . . . . . . 142
WLAN settings (Wizard) . . . . . . . . . . . . . . . . . . . . . 139
WLAN subnet mask (Parameter) . . . . . . . . . . . . . . . 143

U

Upper range limit (Parameter) . . . . . . . . . . . . . . . . . . 59
Upper range value output (Parameter) . . . . . . . . . . . . 78
User name (Parameter) . . . . . . . . . . . . . . . . . . . . . . 141
User role (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 12

V

Value (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . . . 115
Value 1 display (Parameter) . . . . . . . . . . . . . . . . . . . 17
Value 2 display (Parameter) . . . . . . . . . . . . . . . . . . . 19
Value 3 display (Parameter) . . . . . . . . . . . . . . . . . . . 20
Value 4 display (Parameter) . . . . . . . . . . . . . . . . . . . 22
Value 5 display (Parameter) . . . . . . . . . . . . . . . . . . . 23
Value 6 display (Parameter) . . . . . . . . . . . . . . . . . . . 25
Value 7 display (Parameter) . . . . . . . . . . . . . . . . . . . 26
Value 8 display (Parameter) . . . . . . . . . . . . . . . . . . . 28
Value current input 1 to n (Parameter) . . . . . . . . . . . 181
Value current output 1 to n (Submenu) . . . . . . . . . . . 51
Value status input (Parameter) . . . . . . . . . . . . . . 50, 73
Value status input 1 to n (Submenu) . . . . . . . . . . . . . 50
Volume flow unit (Parameter) . . . . . . . . . . . . . . . . . . 56

W

Web server (Submenu) . . . . . . . . . . . . . . . . . . . . . . 135
Web server functionality (Parameter) . . . . . . . . . . . 138
Web server language (Parameter) . . . . . . . . . . . . . . 136
WizardDefine access code . . . . . . . . . . . . . . . . . . . . . . . . 41
WLAN settings . . . . . . . . . . . . . . . . . . . . . . . . . . 139
WLAN (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . . 140
WLAN IP address (Parameter) . . . . . . . . . . . . . . . . . 142
WLAN MAC address (Parameter) . . . . . . . . . . . . . . 142
WLAN mode (Parameter) . . . . . . . . . . . . . . . . . . . . 140
194

Endress+Hauser

*71637048*
71637048

www.addresses.endress.com