EA01144D/06/A2/07.23-00
71616945
2024-01-01

Products

Solutions

Services

Installation Instructions
Replacing the display module
Proline 300, 500, 500-digital, DKX001–

EA01144D

Replacing the display module
Proline 300, 500, 500-digital, DKX001–
Table of contents
1

Overview of spare part sets . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4

2

Designated use . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4

3

Personnel authorized to carry out repairs . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 5

4

Safety instructions . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 5

5Symbols . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6

6Tool list . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 7

7Promag, Promass, Prosonic Flow 300, 500, t-mass, Teqwave M 300 . . . . . . . . . . . . . . . . . 7

8Promag, Promass, Prosonic Flow, t-mass, Teqwave M 300 . . . . . . . . . . . . . . . . . . . . . . . . . . 8

9Promag, Promass, Prosonic Flow, t-mass, Teqwave M 500-digital . . . . . . . . . . . . . . . . . . . . 9

10Disposal . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 11

Endress+Hauser

3

EA01144D

1

Overview of spare part sets

The Installation Instructions apply to the following spare part sets:
Order structure

Device components

XPD0031–

1 × display module, 300, 500, DKX001

XPD0039–

1 × display module, 300, 500, OEM• The order number of the spare part set (on the product label on the package) can differ
from the production number (on the label directly on the spare part)!
• You can find the order number of the relevant spare part set by entering the
production number of the spare part in the spare part search tool.
• We recommend that you keep the Installation Instructions and packaging together at
all times.

2

Designated use

The spare part sets and Installation Instructions are used to replace a faulty unit with afunctioning unit of the same type. Only original parts from Endress+Hauser may be used. Only
spare parts kits designed by Endress+Hauser for the measuring device can be used at any
time.
Inspection is performed using the W@M Device Viewer. The procedure involved is describedas follows.
A spare parts overview is located in the interior of some measuring devices. If the spareparts kit is specified on this list, no inspection is required.

1.

2.

www.endress.com/deviceviewer

Ser. No.: 12345

3.

4.

Order code=

Description?

1.

www.endress.com/deviceviewer

2.

Enter serial number (ser. no.), search for productdata and click Spare Parts.

3.

All spare parts for the measuring device aredisplayed.

4.

Determine the order number of the spare parts kit.

spare parts kit may be used only if the order Thenumber of the spare parts kit corresponds to anorder number on the spare parts list.

4

Endress+Hauser

EA01144D

3

Personnel authorized to carry out repairs

Authorization to carry out repairs depends on the measuring device's approval type. The tablebelow shows the authorized group of people in each case.
Whoever carries out the repairs has full responsibility to ensure that work is carried out
safely and to the required quality standard. He/she must also guarantee the safety of thedevice following repair.

1)

4

Measuring device approval

Personnel authorized to perform repairs 1)

Without approval

1, 2, 3

With approval (e.g. IECEx)

1, 2, 3For custody transfer

4

1 = Qualified specialist on customer side, 2 = Service technician authorized by Endress+Hauser ,3 = Endress+Hauser (return measuring device to manufacturer)
4 = Check with local approval center if installation/modification must be performed under supervision.

Safety instructions• Check whether the spare part matches the identification labeling on the measuring device,
as described on the cover page.
• The spare part set and the Installation Instructions are used to replace a defective unit witha functioning unit of the same type.
Only use original parts from Endress+Hauser.
• Comply with national regulations regarding mounting, electrical installation,commissioning, maintenance and repair.
• The following requirements must be met with regard to specialized technical staff for themounting, electrical installation, commissioning, maintenance and repair of the measuring
devices:• Specialized technical staff must be trained in instrument safety.
• They must be familiar with the individual operating conditions of the devices.
• In the case of Ex-certified measuring devices, they must also be trained in explosion
protection.
• The measuring device is energized! Risk of fatal injury from electric shock. Open themeasuring device only when the device is de-energized.
• For measuring devices intended for use in hazardous locations, please observe theguidelines in the Ex documentation (XA).
• For measuring devices in safety-related applications in accordance with IEC 61508 or
IEC 61511: following repair, re-commission the device in accordance with the Operating
Instructions. Document the repair.
• Before removing the device: set the process to a safe state and purge the pipe of dangerous
process substances.
• Danger of burns due to heated surfaces! Before commencing work: allow the system andmeasuring device to cool down to a touchable temperature.
Endress+Hauser

5

EA01144D• In the case of devices in custody transfer, the custody transfer status no longer applies oncethe seal has been removed.
• The Operating Instructions for the device must be followed.
• Risk of damaging the electronic components! Ensure you have a working environment
protected from electrostatic discharge.
• After removing the electronics compartment cover: risk of electrical shock due to missing
touch protection!
Turn the measuring device off before removing internal covers.
• Modifications to the measuring device are not permitted.
• Only open the housing for a brief period. Avoid foreign objects, moisture or dirt entering the
housing.
• Replace defective seals only with original seals from Endress+Hauser.
• If threads are defective the measuring device must be repaired.
• Threads (e.g. of the electronics compartment cover and connection compartment cover)must be lubricated if an abrasion-proof dry lubricant is not available. Use acid-free, nonhardening lubricant.
• If, during repair work, spacing is reduced or the dielectric strength of the measuring device
cannot be guaranteed, perform a test on completion of the work (e.g. high-voltage test inaccordance with the manufacturer's instructions).
• Service plug:• Do not connect in potentially explosive atmospheres.
• Only connect to Endress+Hauser service devices.
• Observe the instructions for transporting and returning the device outlined in the Operating
Instructions.
Contact Endress+Hauser Service if you have questions: www.addresses.endress.com

5

Symbols

5.1

Symbols for certain types of information

SymbolMeaning
Permitted
Procedures, processes or actions that are permitted.
Forbidden
Procedures, processes or actions that are forbidden.
Tip
Indicates additional information.

1. , 2. , 3. …

6

Series of steps

Endress+Hauser

EA01144D

6

Tool list

Allen key

Torx screwdriver

3 mm

T20

7

Promag, Promass, Prosonic Flow 300, 500, t-mass,
Teqwave M 300

7.1

Aluminum transmitter housing, non-Ex, Ex, stainless cast transmitter housing, Ex

7.1.1

Replacing the standard display module, WLAN, display module carrier

1→

2→

1.
2.
n
f ne
öf

un
te r

Nich
Sp t unPowe
an ter r

n
fne
öf

un
te r

Do nu
en not ng öff
op ne
Neergizeenso pas d wh n
us ouenten vri
sio r
n

Nicht

I/O

Nicht

3.

ES

C

+
E

1.

e

ar

3 mm

1.

2.

ar

e

–

3.

3

7.1.2

Reassembling the standard display module, WLAN, display module carrier

Reassembly is carried out in reverse order.

Endress+Hauser

7

EA01144D

8

Promag, Promass, Prosonic Flow, t-mass,
Teqwave M 300

8.1

Remote display and operating module DKX001

8.1.1

Removing the display module

1→

2

1.

3.

2.
E
C

S

2.
1.

8.1.2

3 mm

Reassembling the display module

Reassembly is carried out in reverse order.

8

Endress+Hauser

EA01144D

9

Promag, Promass, Prosonic Flow, t-mass,
Teqwave M 500-digital

9.1

Aluminum transmitter housing

9.1.1

Replacing the standard display module, WLAN

1→

1.
T20

2.
2→

1.

ES

-

C

+

-

2.

3

1.

ES

-

C

+

-

2.

Endress+Hauser

9EA01144D

9.1.2

Reassembling the standard display module, WLAN, aluminum transmitter
housing

Reassembly is carried out in reverse order.

9.2

Polycarbonate transmitter housing

9.2.1

Replacing the standard display module, WLAN

1→

1.
T20

2.
2→

1.

ES

-

C

+

-

2.

3

1.

ES

-

C

+

-

2.

10

Endress+Hauser

EA01144D

9.2.2

Reassembling the standard display module, WLAN, polycarbonate transmitter
housing

Reassembly is carried out in reverse order.

10

Disposal

If required by the Directive 2012/19/EU on waste electrical and electronic equipment(WEEE), the product is marked with the depicted symbol in order to minimize the disposalof WEEE as unsorted municipal waste. Do not dispose of products bearing this marking as
unsorted municipal waste. Instead, return them to the manufacturer for disposal under theapplicable conditions.

Endress+Hauser

11

EA01144D

Austausch Anzeigemodul
Proline 300, 500, 500-digital, DKX001–
Inhaltsverzeichnis
1Übersicht Ersatzteilsets . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 14

2

Bestimmungsgemäße Verwendung . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 14

3

Reparaturberechtigte Personen . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 15

4

Sicherheitshinweise . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 15

5Verwendete Symbole . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 16

6

Werkzeugliste . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 17

7

Promag, Promass, Prosonic Flow 300, 500, t-mass, Teqwave M 300 . . . . . . . . . . . . . . . . 17

8

Promag, Promass, Prosonic Flow, t-mass, Teqwave M 300 . . . . . . . . . . . . . . . . . . . . . . . . . 18

9

Promag, Promass, Prosonic Flow, t-mass, Teqwave M 500-digital . . . . . . . . . . . . . . . . . . 19

10

Entsorgung . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 21

Endress+Hauser

13

EA01144D

1Übersicht Ersatzteilsets

Die Einbauanleitung ist für folgende Ersatzteilsets gültig:
Bestellstruktur

Gerätekomponente

XPD0031–

1 × Anzeigemodul, 300, 500, DKX001

XPD0039–

1 × Anzeigemodul, 300, 500, OEM• Die Bestellnummer des Ersatzteilsets (auf dem Produktaufkleber der Verpackung)kann sich von der Produktionsnummer (auf dem Aufkleber direkt auf dem Ersatzteil)unterscheiden!
• Durch Eingabe der Produktionsnummer des Ersatzteiles im Ersatzteilfindetool kanndie Bestellnummer des entsprechenden Ersatzteilsets ermittelt werden.
• Wir empfehlen Einbauanleitung und Verpackung immer zusammen aufzubewahren.

2

Bestimmungsgemäße Verwendung

Ersatzteilset und Einbauanleitung dienen dazu, eine defekte Einheit gegen einefunktionierende Einheit des gleichen Typs zu ersetzen. Es dürfen nur Originalteile vonEndress+Hauser verwendet werden. Grundsätzlich dürfen nur Ersatzteilsets verwendet
werden, die von Endress+Hauser für das Messgerät vorgesehen sind.
Die Überprüfung ist via W@M Device Viewer durchzuführen, die Vorgehensweise dazu ist
nachfolgend beschrieben.
Bei einigen Messgeräten befindet sich im Inneren des Gerätes eine Ersatzteilübersicht.
Ist das Ersatzteilset dort aufgelistet, entfällt die Überprüfung.

1.

2.

www.endress.com/deviceviewer

Ser. No.: 12345

3.

4.

14

Order code=

Description?

1.

www.endress.com/deviceviewer

2.

Seriennummer (Ser. No.) eingeben, Produktdaten
suchen und auf Ersatzteile klicken.

3.

Anzeige aller Ersatzteile zum Messgerät.

4.

Die Bestellnummer des Ersatzteilsets ermitteln.

Nur wenn die Bestellnummer des Ersatzteilsets mit
 einer
Bestellnummer in der Ersatzteillisteübereinstimmt, darf das Ersatzteilset verwendet
werden.

Endress+Hauser

EA01144D

3

Reparaturberechtigte Personen

Die Berechtigung zur Durchführung einer Reparatur ist von der Zulassung des Messgeräts
abhängig. Die Tabelle zeigt den jeweils berechtigten Personenkreis.
Die Person, die eine Reparatur vornimmt, übernimmt die Verantwortung für die
Sicherheit während der Arbeiten, die Qualität der Ausführung und die Sicherheit desGeräts nach der Reparatur.

1)

4

Zulassung des Messgeräts

Reparaturberechtigter Personenkreis 1)

Ohne Zulassung

1, 2, 3Mit Zulassung (z.B. IECEx)

1, 2, 3Bei eichfähigem Verkehr

4

1 = Ausgebildete Fachkraft des Kunden, 2 = Von Endress+Hauser autorisierter Servicetechniker,
3 = Endress+Hauser (Messgerät an Hersteller zurücksenden)
4 = Mit der lokalen Zulassungsstelle prüfen, ob ein Ein-/Umbau unter Aufsicht erfolgen muss.

Sicherheitshinweise• Prüfen, ob das vorliegende Ersatzteil zur Kennzeichnung auf dem Messgerät passt, wie auf
der Titelseite beschrieben.
• Ersatzteilset und Einbauanleitung dienen dazu, eine defekte Einheit gegen einefunktionierende Einheit des gleichen Typs zu ersetzen.
Nur Originalteile von Endress+Hauser verwenden.
• Nationale Vorschriften bezüglich der Montage, elektrischen Installation, Inbetriebnahme,
Wartung und Reparatur einhalten.
• Folgende Anforderungen an das Fachpersonal für Montage, elektrische Installation,
Inbetriebnahme, Wartung und Reparatur der Messgeräte müssen erfüllt sein:• In Gerätesicherheit ausgebildet.
• Mit den jeweiligen Einsatzbedingungen der Geräte vertraut.
• Bei Ex-zertifizierten Messgeräten: zusätzlich im Explosionsschutz ausgebildet.
• Messgerät unter Spannung! Lebensgefahr durch Stromschlag. Messgerät nur im
spannungslosen Zustand öffnen.
• Bei Messgeräten für den explosionsgefährdeten Bereich: Hinweise in der ExDokumentation (XA) beachten.
• Bei Messgeräten in sicherheitstechnischen Applikationen gemäß IEC 61508 bzw.
IEC 61511: Nach Reparatur Neuinbetriebnahme gemäß Betriebsanleitung durchführen.
Reparatur dokumentieren.
• Vor einem Geräteausbau: Prozess in sicheren Zustand bringen und Leitung von gefährlichen
Prozessstoffen befreien.
• Verbrennungsgefahr durch heiße Oberflächen! Vor Arbeitsbeginn: Anlage und Messgerät
auf berührungssichere Temperatur abkühlen.
• Bei Messgeräten im abrechnungspflichtigen Verkehr: Nach Entfernen der Plombe ist der
geeichte Zustand aufgehoben.
• Die Betriebsanleitung zum Messgerät ist zu beachten.
Endress+Hauser

15

EA01144D• Beschädigungsgefahr elektronischer Bauteile! Eine ESD-geschützte Arbeitsumgebung
herstellen.
• Nach Entfernen der Elektronikabdeckung: Stromschlaggefahr durch aufgehobenen
Berührungsschutz!
Messgerät ausschalten, bevor interne Abdeckungen entfernt werden.
• Änderungen am Messgerät sind nicht zulässig.
• Gehäuse nur kurzzeitig öffnen. Eindringen von Fremdkörpern, Feuchtigkeit oder
Verunreinigung vermeiden.
• Defekte Dichtungen nur durch Original-Dichtungen von Endress+Hauser ersetzen.
• Defekte Gewinde erfordern eine Instandsetzung des Messgeräts.
• Gewinde (z.B. von Elektronikraum- und Anschlussraumdeckel) müssen geschmiert sein,
sofern keine abriebfeste Trockenschmierung vorhanden ist. Säurefreies, nicht härtendes
Fett verwenden.
• Wenn bei den Reparaturarbeiten Abstände reduziert oder die Spannungsfestigkeit desMessgeräts nicht sichergestellt werden kann: Prüfung nach Abschluss der Arbeiten
durchführen (z.B. Hochspannungstest gemäß Herstellerangaben).
• Servicestecker:• Nicht in explosionsfähiger Atmosphäre anschließen.
• Nur an Servicegeräte von Endress+Hauser anschließen.
• Die in der Betriebsanleitung aufgeführten Hinweise zum Transport und zur Rücksendung
beachten.
Bei Fragen Endress+Hauser Service kontaktieren: www.addresses.endress.com

5

Verwendete Symbole

5.1

Symbole für Informationstypen

SymbolBedeutung
Erlaubt
Abläufe, Prozesse oder Handlungen, die erlaubt sind.
Verboten
Abläufe, Prozesse oder Handlungen, die verboten sind.
Tipp
Kennzeichnet zusätzliche Informationen.

1. , 2. , 3. …

16

Handlungsschritte

Endress+Hauser

EA01144D

6

Werkzeugliste

Innensechskant-Torxschlüssel

Schraubenzieher

3 mm

T20

7

Promag, Promass, Prosonic Flow 300, 500, t-mass,
Teqwave M 300

7.1

Messumformergehäuse Aluminium Ex-frei, Ex, Messumformergehäuse Guss rostfrei Ex

7.1.1

Austausch Anzeigemodul Standard, WLAN, Anzeigemodulträger

1→

2→

1.
2.
n
f ne
öf

un
te r

un
te r

Nicht

I/O

Nicht

3.

n
fne
öf

Nich
Sp t unPowean
Do nu ter r
en not ng öff
op ne
Neergizeenso pas d wh n
us ouenten vri
sio r
n

ES

C

+
E

e

ar

3 mm

1.

2.

ar

e

–

1.

3.

3

7.1.2

Zusammenbau Anzeigemodul Standard, WLAN, Anzeigemodulträger

Der Zusammenbau erfolgt in umgekehrter Reihenfolge.
Endress+Hauser

17

EA01144D

8

Promag, Promass, Prosonic Flow, t-mass,
Teqwave M 300

8.1

Abgesetztes Anzeige-und Bedienmodul DKX001

8.1.1

Ausbau Anzeigemodul

1→

2

1.

3.

2.
E
C

S

2.
1.

8.1.2

3 mm

Zusammenbau Anzeigemodul

Der Zusammenbau erfolgt in umgekehrter Reihenfolge.

18

Endress+Hauser

EA01144D

9

Promag, Promass, Prosonic Flow, t-mass,
Teqwave M 500-digital

9.1

Messumformergehäuse Aluminium

9.1.1

Austausch Anzeigemodul Standard, WLAN

1→

1.
T20

2.
2→

1.

ES

-

C

+

-

2.

3

1.

ES

-

C

+

-

2.

Endress+Hauser

19

EA01144D

9.1.2

Zusammenbau Anzeigemodul Standard, WLAN, Messumformergehäuse
Aluminium

Der Zusammenbau erfolgt in umgekehrter Reihenfolge.

9.2

Messumformergehäuse Polycarbonat

9.2.1

Austausch Anzeigemodul Standard, WLAN

1→

1.
T20

2.
2→

1.

ES

-

C

+

-

2.

3

1.

ES

-

C

+

-

2.

20

Endress+Hauser

EA01144D

9.2.2

Zusammenbau Anzeigemodul Standard, WLAN, Messumformergehäuse
Polycarbonat

Der Zusammenbau erfolgt in umgekehrter Reihenfolge.

10

Entsorgung

Gemäß der Richtlinie 2012/19/EU über Elektro- und Elektronik-Altgeräte (WEEE) ist das
Produkt mit dem abgebildeten Symbol gekennzeichnet, um die Entsorgung von WEEE als
unsortierten Hausmüll zu minimieren. Gekennzeichnete Produkte nicht als unsortierter
Hausmüll entsorgen, sondern zu den gültigen Bedingungen an den Hersteller zurückgeben.

Endress+Hauser21

*71616945*
71616945

www.addresses.endress.com