BA02320D/06/EN/01.23-00
71637042
2024-03-15

Products

Solutions

Services

Valid as of version
01.00.zz (Device firmware)

Operating Instructions
Proline Teqwave MW 300
Total solids measurement via microwave transmission
HARTProline Teqwave MW 300 HART

• Make sure the document is stored in a safe place such that it is always available whenworking on or with the device.
• To avoid danger to individuals or the facility, read the "Basic safety instructions" section
carefully, as well as all other safety instructions in the document that are specific toworking procedures.
• The manufacturer reserves the right to modify technical data without prior notice. Your
Endress+Hauser sales organization will supply you with current information and updates
to this manual.

2

Endress+Hauser

Proline Teqwave MW 300 HART

Table of contents

Table of contents
1

About this document . . . . . . . . . . . . . . . . 6

1.1
1.2

1.4

Document function . . . . . . . . . . . . . . . . . . . . . 6
Symbols . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6
1.2.1 Safety symbols . . . . . . . . . . . . . . . . . . 6
1.2.2 Electrical symbols . . . . . . . . . . . . . . . . 6
1.2.3 Communication-specific symbols . . . . . 6
1.2.4 Tool symbols . . . . . . . . . . . . . . . . . . . . 7
1.2.5 Symbols forcertain types of information . . . . . . . . . 7
1.2.6 Symbols in graphics . . . . . . . . . . . . . . . 7
Documentation . . . . . . . . . . . . . . . . . . . . . . . . 8
1.3.1 Document function . . . . . . . . . . . . . . . 8
Registered trademarks . . . . . . . . . . . . . . . . . . . 82

Safety instructions . . . . . . . . . . . . . . . . . . 9

2.1
2.2
2.3
2.4
2.5
2.6
2.7

Requirements for the personnel . . . . . . . . . . . . 9
Intended use . . . . . . . . . . . . . . . . . . . . . . . . . . 9
Workplace safety . . . . . . . . . . . . . . . . . . . . . . 10
Operational safety . . . . . . . . . . . . . . . . . . . . . 10
Product safety . . . . . . . . . . . . . . . . . . . . . . . . 10
IT security . . . . . . . . . . . . . . . . . . . . . . . . . . . 10
Device-specific IT security . . . . . . . . . . . . . . . . 11
2.7.1 Protecting access via hardware write
protection . . . . . . . . . . . . . . . . . . . . . 11
2.7.2 Protecting access via a password . . . . 11
2.7.3 Access via web server . . . . . . . . . . . . 12
2.7.4 Access via service interface (CDIRJ45) . . . . . . . . . . . . . . . . . . . . . . . . 12

1.3

3

Product description . . . . . . . . . . . . . . . . 13

3.1

Product design . . . . . . . . . . . . . . . . . . . . . . . . 13

4

Incoming acceptance and productidentification . . . . . . . . . . . . . . . . . . . . . 14

4.1
4.2

Incoming acceptance . . . . . . . . . . . . . . . . . . . 14
Product identification . . . . . . . . . . . . . . . . . . . 15
4.2.1 Transmitter nameplate . . . . . . . . . . . 16
4.2.2 Sensor nameplate . . . . . . . . . . . . . . . 17
4.2.3 Symbols on the device . . . . . . . . . . . . 18

5

Storage and transport . . . . . . . . . . . . . 19

5.1
5.2
5.3

Storage conditions . . . . . . . . . . . . . . . . . . . . . 19
Transporting the product . . . . . . . . . . . . . . . . 19
5.2.1 Measuring devices with lifting lugs . . 19
Packaging disposal . . . . . . . . . . . . . . . . . . . . . 20

6

Mounting procedure . . . . . . . . . . . . . . . 21

6.1

Mounting requirements . . . . . . . . . . . . . . . . . 21
6.1.1 Mounting position . . . . . . . . . . . . . . . 21
6.1.2 Environmental and process
requirements . . . . . . . . . . . . . . . . . . 25

Endress+Hauser

6.3

6.1.3 Special mounting instructions . . . . . .
Mounting the measuring device . . . . . . . . . . .
6.2.1 Preparing the measuring device . . . . .
6.2.2 Mounting the sensor . . . . . . . . . . . . .
6.2.3 Turning the transmitter housing . . . .
6.2.4 Turning the display module . . . . . . . .
Post-mounting check . . . . . . . . . . . . . . . . . . .

7Electrical connection . . . . . . . . . . . . . . 33

7.1
7.2

7.6
7.7

Electrical safety . . . . . . . . . . . . . . . . . . . . . . . 33
Connecting requirements . . . . . . . . . . . . . . . . 33
7.2.1 Required tools . . . . . . . . . . . . . . . . . . 33
7.2.2 Requirements for connection cable . . . 33
7.2.3 Terminal assignment . . . . . . . . . . . . . 35
7.2.4 Preparing the measuring device . . . . . 35
Connecting the measuring device . . . . . . . . . . 35
7.3.1 Connecting the transmitter . . . . . . . . 36
7.3.2 Connecting the remote display andoperating module DKX001 . . . . . . . . 39
Potential equalization . . . . . . . . . . . . . . . . . . 39
7.4.1 Requirements . . . . . . . . . . . . . . . . . . 39
Special connection instructions . . . . . . . . . . . . 40
7.5.1 Connection examples . . . . . . . . . . . . . 40
Ensuring the degree of protection . . . . . . . . . . 44
Post-connection check . . . . . . . . . . . . . . . . . . 45

8Operation options . . . . . . . . . . . . . . . . . 46

8.1
8.2

Overview of operation options . . . . . . . . . . . . 46
Structure and function of the operating
menu . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 47
8.2.1 Structure of the operating menu . . . . 47
8.2.2 Operating philosophy . . . . . . . . . . . . 48
Access to operating menu via local display . . . 49
8.3.1 Operational display . . . . . . . . . . . . . . 49
8.3.2 Navigation view . . . . . . . . . . . . . . . . 51
8.3.3 Editing view . . . . . . . . . . . . . . . . . . . 53
8.3.4 Operating elements . . . . . . . . . . . . . . 55
8.3.5 Opening the context menu . . . . . . . . . 55
8.3.6 Navigating and selecting from list . . . 57
8.3.7 Calling up help text . . . . . . . . . . . . . . 57
8.3.8 Changing the parameters . . . . . . . . . 58
8.3.9 User roles and related accessauthorization . . . . . . . . . . . . . . . . . . 58
8.3.10 Disabling write protection via accesscode . . . . . . . . . . . . . . . . . . . . . . . . . 59
8.3.11 Enabling and disabling the keypadlock . . . . . . . . . . . . . . . . . . . . . . . . . 59
Access to operating menu via web browser . . . 60
8.4.1 Function range . . . . . . . . . . . . . . . . . 60
8.4.2 Requirements . . . . . . . . . . . . . . . . . . 60
8.4.3 Connecting the device . . . . . . . . . . . . 61
8.4.4 Logging on . . . . . . . . . . . . . . . . . . . . 63
8.4.5 User interface . . . . . . . . . . . . . . . . . . 64
8.4.6 Disabling the Web server . . . . . . . . . . 65

6.2

7.3

7.4
7.5

8.3

8.4

26
27
27
27
29
30
32

3

Table of contents

8.5

8.4.7 Logging out . . . . . . . . . . . . . . . . . . . . 66
Access to the operating menu via theoperating tool . . . . . . . . . . . . . . . . . . . . . . . . 66
8.5.1 Connecting the operating tool . . . . . . 66
8.5.2 Field Xpert SFX350, SFX370 . . . . . . . 70
8.5.3 FieldCare . . . . . . . . . . . . . . . . . . . . . 70
8.5.4 DeviceCare . . . . . . . . . . . . . . . . . . . . 71
8.5.5 AMS Device Manager . . . . . . . . . . . . 72
8.5.6 Field Communicator 475 . . . . . . . . . . 72
8.5.7 SIMATIC PDM . . . . . . . . . . . . . . . . . . 72

9

System integration . . . . . . . . . . . . . . . . 73

9.1

9.3

Overview of device description files . . . . . . . . . 73
9.1.1 Current version data for the device . . . 73
9.1.2 Operating tools . . . . . . . . . . . . . . . . . 74
Measured variables via HART protocol . . . . . . 75
9.2.1 Dynamic variables . . . . . . . . . . . . . . 75
9.2.2 Device variables . . . . . . . . . . . . . . . . . 76
Other settings . . . . . . . . . . . . . . . . . . . . . . . . 77

10

Commissioning . . . . . . . . . . . . . . . . . . . . 80

9.2

10.1 Post-mounting and post-connection check . . . 80
10.2 Switching on the measuring device . . . . . . . . . 80
10.3 Connecting via FieldCare . . . . . . . . . . . . . . . . 80
10.4 Setting the operating language . . . . . . . . . . . . 80
10.5 Configuring the measuring device . . . . . . . . . . 81
10.5.1 Defining the tag name . . . . . . . . . . . . 82
10.5.2 Setting the system units . . . . . . . . . . 83
10.5.3 Displaying the I/O configuration . . . . 84
10.5.4 Configuring the current input . . . . . . 85
10.5.5 Configuring the status input . . . . . . . 86
10.5.6 Configuring the current output . . . . . 87
10.5.7 Configuring the pulse/frequency/
switch output . . . . . . . . . . . . . . . . . . 89
10.5.8 Configuring the relay output . . . . . . . 95
10.5.9 Configuring the local display . . . . . . . 96
10.5.10 "Total solids commissioning" wizard . . 99
10.5.11 "Total solids adjustment" submenu . . . 99
10.6 Advanced settings . . . . . . . . . . . . . . . . . . . . 100
10.6.1 Using the parameter to enter theaccess code . . . . . . . . . . . . . . . . . . . 101
10.6.2 Configuring the totalizer . . . . . . . . . 102
10.6.3 Carrying out additional display
configurations . . . . . . . . . . . . . . . . . 103
10.6.4 WLAN configuration . . . . . . . . . . . . 107
10.6.5 Heartbeat Technology application
package . . . . . . . . . . . . . . . . . . . . . 108
10.6.6 Configuration management . . . . . . . 109
10.6.7 Using parameters for device
administration . . . . . . . . . . . . . . . . 111
10.7 Simulation . . . . . . . . . . . . . . . . . . . . . . . . . . 113
10.8 Protecting settings from unauthorized access 116
10.8.1 Write protection via access code . . . 116
10.8.2 Write protection via write protection
switch . . . . . . . . . . . . . . . . . . . . . . . 117

4

Proline Teqwave MW 300 HART

11

Operation . . . . . . . . . . . . . . . . . . . . . . . 119

11.1 Reading off the device locking status . . . . . . 119
11.2 Adjusting the operating language . . . . . . . . . 119
11.3 Configuring the display . . . . . . . . . . . . . . . . 119
11.4 Adapting the measuring device to the process
conditions . . . . . . . . . . . . . . . . . . . . . . . . . . 119
11.5 Reading off measured values . . . . . . . . . . . . 120
11.5.1 "Process variables" submenu . . . . . . . 120
11.5.2 "Input values" submenu . . . . . . . . . . 121
11.5.3 Output values . . . . . . . . . . . . . . . . . 122
11.5.4 "Totalizer" submenu . . . . . . . . . . . . . 124
11.6 Performing a totalizer reset . . . . . . . . . . . . . 125
11.6.1 Function scope of "Control Totalizer"
parameter . . . . . . . . . . . . . . . . . . . . 125
11.6.2 Function range of "Reset all
totalizers" parameter . . . . . . . . . . . . 126
11.7 Displaying the measured value history . . . . . 127
11.8 Adjusting the measured value with the aid of
wizards . . . . . . . . . . . . . . . . . . . . . . . . . . . . 130
11.8.1 Performing the basic settings for the
adjustment . . . . . . . . . . . . . . . . . . . 130
11.8.2 Adjusting the measured value basedon the reference value . . . . . . . . . . . 130
11.8.3 Calling up wizards . . . . . . . . . . . . . . 132

12

Diagnosis and troubleshooting . . . . 134

12.1
12.2

General troubleshooting . . . . . . . . . . . . . . . . 134
Diagnostic information via LEDs . . . . . . . . . . 136
12.2.1 Transmitter . . . . . . . . . . . . . . . . . . . 136
12.3 Diagnostic information on local display . . . . . 137
12.3.1 Diagnostic message . . . . . . . . . . . . . 137
12.3.2 Calling up remedial measures . . . . . 139
12.4 Diagnostic information in the web browser . . 140
12.4.1 Diagnostic options . . . . . . . . . . . . . . 140
12.4.2 Calling up remedy information . . . . 141
12.5 Diagnostic information in FieldCare or
DeviceCare . . . . . . . . . . . . . . . . . . . . . . . . . 142
12.5.1 Diagnostic options . . . . . . . . . . . . . . 142
12.5.2 Calling up remedy information . . . . 143
12.6 Adapting the diagnostic information . . . . . . 144
12.6.1 Adapting the diagnostic behavior . . . 144
12.6.2 Adapting the status signal . . . . . . . . 144
12.7 Overview of diagnostic information . . . . . . . 146
12.8 Pending diagnostic events . . . . . . . . . . . . . . 150
12.9 Diagnostics list . . . . . . . . . . . . . . . . . . . . . . 150
12.10 Event logbook . . . . . . . . . . . . . . . . . . . . . . . 152
12.10.1 Reading out the event logbook . . . . . 152
12.10.2 Filtering the event logbook . . . . . . . 152
12.10.3 Overview of information events . . . . 153
12.11 Resetting the measuring device . . . . . . . . . . 155
12.11.1 Function range of "Device reset"
parameter . . . . . . . . . . . . . . . . . . . . 155
12.12 Device information . . . . . . . . . . . . . . . . . . . 156
12.13 Firmware history . . . . . . . . . . . . . . . . . . . . . 158

Endress+Hauser

Proline Teqwave MW 300 HART

13

Maintenance . . . . . . . . . . . . . . . . . . . . 159

13.1
13.2

Maintenance work . . . . . . . . . . . . . . . . . . . . 159
13.1.1 Exterior cleaning . . . . . . . . . . . . . . . 159
Endress+Hauser services . . . . . . . . . . . . . . . 159

14

Repair . . . . . . . . . . . . . . . . . . . . . . . . . . . 160

Table of contents

14.1

General notes . . . . . . . . . . . . . . . . . . . . . . . 160
14.1.1 Repair and conversion concept . . . . . 160
14.1.2 Notes for repair and conversion . . . . 160
14.2 Spare parts . . . . . . . . . . . . . . . . . . . . . . . . . 160
14.3 Endress+Hauser services . . . . . . . . . . . . . . . 160
14.4 Return . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 160
14.5 Disposal . . . . . . . . . . . . . . . . . . . . . . . . . . . 161
14.5.1 Removing the measuring device . . . . 161
14.5.2 Disposing of the measuring device . . 161

15

Accessories . . . . . . . . . . . . . . . . . . . . . . 162

15.1

Device-specific accessories . . . . . . . . . . . . . . 162
15.1.1 For the transmitter . . . . . . . . . . . . . 162
15.1.2 For the sensor . . . . . . . . . . . . . . . . . 163
15.2 Communication-specific accessories . . . . . . . 163
15.3 Service-specific accessories . . . . . . . . . . . . . . 164
15.4 System components . . . . . . . . . . . . . . . . . . . 165

16

Technical data . . . . . . . . . . . . . . . . . . . 166

16.1 Application . . . . . . . . . . . . . . . . . . . . . . . . . 166
16.2 Function and system design . . . . . . . . . . . . . 166
16.3 Input . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 166
16.4 Output . . . . . . . . . . . . . . . . . . . . . . . . . . . . 169
16.5 Power supply . . . . . . . . . . . . . . . . . . . . . . . . 173
16.6 Performance characteristics . . . . . . . . . . . . . 175
16.7 Mounting . . . . . . . . . . . . . . . . . . . . . . . . . . 176
16.8 Environment . . . . . . . . . . . . . . . . . . . . . . . . 176
16.9 Process . . . . . . . . . . . . . . . . . . . . . . . . . . . . 178
16.10 Mechanical construction . . . . . . . . . . . . . . . 178
16.11 Display and user interface . . . . . . . . . . . . . . 180
16.12 Certificates and approvals . . . . . . . . . . . . . . 188
16.13 Application packages . . . . . . . . . . . . . . . . . . 190
16.14 Supplemental documentation . . . . . . . . . . . . 191

Index . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 193

Endress+Hauser

5About this document

Proline Teqwave MW 300 HART

1

About this document

1.1

Document function

These Operating Instructions contain all the information required in the various life cycle
phases of the device: from product identification, incoming acceptance and storage, toinstallation, connection, operation and commissioning, through to troubleshooting,
maintenance and disposal.

1.2

Symbols

1.2.1

Safety symbols

DANGER

This symbol alerts you to a dangerous situation. Failure to avoid this situation will result inserious or fatal injury.
WARNING

This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inserious or fatal injury.
CAUTION

This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inminor or medium injury.
NOTICE

This symbol contains information on procedures and other facts which do not result inpersonal injury.

1.2.2

Electrical symbols
SymbolMeaning
Direct current
Alternating current
Direct current and alternating current
Ground connection
A grounded terminal which, as far as the operator is concerned, is grounded via agrounding system.
Potential equalization connection (PE: protective earth)Ground terminals that must be connected to ground prior to establishing any otherconnections.
The ground terminals are located on the interior and exterior of the device:• Interior ground terminal: potential equalization is connected to the supply network.
• Exterior ground terminal: device is connected to the plant grounding system.

1.2.3

Communication-specific symbols
SymbolMeaning
Wireless Local Area Network (WLAN)
Communication via a wireless, local network.
LED
Light emitting diode is off.

6

Endress+Hauser

Proline Teqwave MW 300 HART

About this document

SymbolMeaning
LED
Light emitting diode is on.
LED
Light emitting diode is flashing.

1.2.4

Tool symbols
SymbolMeaning
Flat-blade screwdriver
Allen key
Open-ended wrench

1.2.5

Symbols for certain types of information
SymbolMeaning
Permitted
Procedures, processes or actions that are permitted.
Preferred
Procedures, processes or actions that are preferred.
Forbidden
Procedures, processes or actions that are forbidden.
Tip
Indicates additional information.
Reference to documentation
Reference to pageA

Reference to graphic
Notice or individual step to be observed
1. , 2. , 3. …

Series of steps
Result of a step
Help in the event of a problem
Visual inspection

1.2.6

Symbols in graphics
SymbolMeaning

1, 2, 3, ...

Item numbers

1. , 2. , 3. , …

Series of steps

A, B, C, ...

Views

A-A, B-B, C-C, ...

Sections

Endress+Hauser

Hazardous area

7

About this document

Proline Teqwave MW 300 HART

Symbol.

Meaning
Safe area (non-hazardous area)
Flow direction

1.3

Documentation
For an overview of the scope of the associated Technical Documentation, refer to thefollowing:• Device Viewer (www.endress.com/deviceviewer): Enter the serial number from thenameplate• Endress+Hauser Operations app: Enter serial number from nameplate or scanmatrix code on nameplate.

1.3.1

Document function

The following documentation may be available depending on the version ordered:
Document type

Purpose and content of the document

Technical Information (TI)

Planning aid for your device
The document contains all the technical data on the device and provides
an overview of the accessories and other products that can be ordered forthe device.

Brief Operating Instructions (KA)

Guide that takes you quickly to the 1st measured value
The Brief Operating Instructions contain all the essential informationfrom incoming acceptance to initial commissioning.

Operating Instructions (BA)Your reference document
The Operating Instructions contain all the information that is required in
the various phases of the life cycle of the device: from productidentification, incoming acceptance and storage, to mounting,connection, operation and commissioning through to troubleshooting,
maintenance and disposal.

Description of Device Parameters(GP)

Reference for your parameters
The document provides a detailed explanation of each individual
parameter. The description is aimed at those who work with the device
over the entire life cycle and perform specific configurations.

Safety Instructions (XA)

Depending on the approval, safety instructions for electrical equipment inhazardous areas are also supplied with the device. The Safety Instructions
are an integral part of the Operating Instructions.
Information on the Safety Instructions (XA) relevant to the device is provided
on the nameplate.

Supplementary device-dependent
documentation (SD/FY)

1.4

Always comply strictly with the instructions in the relevant
supplementary documentation. The supplementary documentation is an
integral part of the device documentation.

Registered trademarks

HART®
Registered trademark of the FieldComm Group, Austin, Texas USA

8

Endress+Hauser

Proline Teqwave MW 300 HART

Safety instructions

2

Safety instructions

2.1

Requirements for the personnel

The personnel for installation, commissioning, diagnostics and maintenance must fulfill
the following requirements:‣ Trained, qualified specialists must have a relevant qualification for this specific function
and task.
‣ Are authorized by the plant owner/operator.
‣ Are familiar with federal/national regulations.
‣ Before starting work, read and understand the instructions in the manual andsupplementary documentation as well as the certificates (depending on theapplication).
‣ Follow instructions and comply with basic conditions.
The operating personnel must fulfill the following requirements:‣ Are instructed and authorized according to the requirements of the task by the facility'sowner-operator.

‣ Follow the instructions in this manual.

2.2

Intended use

Application and media
The measuring device described in this manual is intended only for solids content
measurement in water-based liquids.
Measuring devices for use in explosive atmospheres are labeled accordingly on the
nameplate.
To ensure that the measuring device is in proper condition during the operation period:‣ Only use the measuring device in full compliance with the data on the nameplate andthe general conditions listed in the Operating Instructions and supplementary
documentation.
‣ Refer to the nameplate to check whether the ordered device can be operated for theintended application in areas requiring specific approvals (e.g. explosion protection,pressure equipment safety).
‣ Use the measuring device only for media to which the process-wetted materials are
sufficiently resistant.
‣ Keep within the specified pressure and temperature range.
‣ Keep within the specified ambient temperature range.
‣ Protect the measuring device permanently against corrosion from environmental
influences.
Incorrect use
Non-designated use can compromise safety. The manufacturer is not liable for damage
caused by improper or non-designated use.

LWARNING
Danger of breakage due to corrosive or abrasive fluids and ambient conditions!
‣ Verify the compatibility of the process fluid with the sensor material.
‣ Ensure the resistance of all fluid-wetted materials in the process.
‣ Keep within the specified pressure and temperature range.

Endress+Hauser

9Safety instructions

Proline Teqwave MW 300 HARTNOTICE
Verification for borderline cases:‣ For special fluids and fluids for cleaning, Endress+Hauser is glad to provide assistancein verifying the corrosion resistance of fluid-wetted materials, but does not accept any
warranty or liability as minute changes in the temperature, concentration or level ofcontamination in the process can alter the corrosion resistance properties.
Residual risks

LCAUTION
Risk of hot or cold burns! The use of media and electronics with high or low
temperatures can produce hot or cold surfaces on the device.
‣ Mount suitable touch protection.
‣ Use suitable protective equipment.

2.3

Workplace safety

When working on and with the device:‣ Wear the required personal protective equipment as per national regulations.

2.4

Operational safety

Damage to the device!

‣ Operate the device in proper technical condition and fail-safe condition only.
‣ The operator is responsible for the interference-free operation of the device.
Modifications to the device
Unauthorized modifications to the device are not permitted and can lead to unforeseeable
dangers!
‣ If modifications are nevertheless required, consult with the manufacturer.
RepairTo ensure continued operational safety and reliability:‣ Carry out repairs on the device only if they are expressly permitted.
‣ Observe federal/national regulations pertaining to the repair of an electrical device.
‣ Use only original spare parts and accessories.

2.5

Product safety

This measuring device is designed in accordance with good engineering practice to meetstate-of-the-art safety requirements, has been tested, and left the factory in a condition inwhich it is safe to operate.
It meets general safety standards and legal requirements. It also complies with the EU
directives listed in the device-specific EU Declaration of Conformity..

2.6

IT security

Our warranty is valid only if the product is installed and used as described in the Operating
Instructions. The product is equipped with security mechanisms to protect it against any
inadvertent changes to the settings.
IT security measures, which provide additional protection for the product and associateddata transfer, must be implemented by the operators themselves in line with their security
standards.

10

Endress+Hauser

Proline Teqwave MW 300 HART

2.7

Safety instructions

Device-specific IT security

The device offers a range of specific functions to support protective measures on theoperator's side. These functions can be configured by the user and guarantee greater inoperation safety if used correctly. The following list provides an overview of the mostimportant functions:
Function/interface

Factory setting

Recommendation

Write protection via hardware write
protection switch →  11

Not enabled

On an individual basis following risk
assessment

Access code(also applies to web server login or
FieldCare connection) →  11

Not enabled(0000)

Assign a customized access code duringcommissioning

WLAN(order option in display module)

EnabledOn an individual basis following risk
assessment

WLAN security mode

Enabled (WPA2PSK)Do not change

WLAN passphrase(Password) →  12

Serial number

Assign an individual WLAN passphrase duringcommissioning

WLAN mode

Access point

On an individual basis following risk
assessment

Web server →  12

EnabledOn an individual basis following risk
assessment

CDI-RJ45 service interface→  12–

On an individual basis following risk
assessment

2.7.1

Protecting access via hardware write protection

Write access to the parameters of the device via the local display, web browser or
operating tool (e.g. FieldCare, DeviceCare) can be disabled via a write protection switch(DIP switch on the main electronics module). When hardware write protection is enabled,
only read access to the parameters is possible.
Hardware write protection is disabled when the device is delivered →  117.

2.7.2

Protecting access via a password

Different passwords are available to protect write access to the device parameters or accessto the device via the WLAN interface.
• User-specific access code
Protect write access to the device parameters via the local display, web browser or
operating tool (e.g. FieldCare, DeviceCare). Access authorization is clearly regulated
through the use of a user-specific access code.
• WLAN passphrase
The network key protects a connection between an operating unit (e.g. notebook ortablet) and the device via the WLAN interface which can be ordered as an option.
• Infrastructure mode
When the device is operated in infrastructure mode, the WLAN passphrase corresponds
to the WLAN passphrase configured on the operator side.
User-specific access code
Write access to the device parameters via the local display, web browser or operating tool(e.g. FieldCare, DeviceCare) can be protected by the modifiable, user-specific access code(→  116).
When the device is delivered, the device does not have an access code and is equivalent to0000 (open).
Endress+Hauser

11

Safety instructions

Proline Teqwave MW 300 HARTWLAN passphrase: Operation as WLAN access point
A connection between an operating unit (e.g. notebook or tablet) and the device via the
WLAN interface (→  68), which can be ordered as an optional extra, is protected by
the network key. The WLAN authentication of the network key complies with theIEEE 802.11 standard.
When the device is delivered, the network key is pre-defined depending on the device. It
can be changed via the WLAN settings submenu in the WLAN passphrase parameter(→  108).
Infrastructure mode
A connection between the device and WLAN access point is protected by means of an SSID
and passphrase on the system side. Please contact the relevant system administrator foraccess.
General notes on the use of passwords• The access code and network key supplied with the device should be changed duringcommissioning for safety reasons.
• Follow the general rules for generating a secure password when defining and managing
the access code and network key.
• The user is responsible for the management and careful handling of the access code andnetwork key.
• For information on configuring the access code or on what to do if you lose thepassword, for example, see "Write protection via access code" →  116.

2.7.3

Access via web server→  60With the integrated web server, the device can be operated and configured via a
web browser. The connection is via the service interface (CDI-RJ45) or the WLAN
interface.
The web server is enabled when the device is delivered. The web server can be disabled via
the Web server functionality parameter if necessary (e.g., after commissioning).
The device and status information can be hidden on the login page. This prevents
unauthorized access to the information.
For detailed information on device parameters, see:"Description of device parameters" document →  191.

2.7.4

Access via service interface (CDI-RJ45)

The device can be connected to a network via the service interface (CDI-RJ45). Devicespecific functions guarantee the secure operation of the device in a network.
The use of relevant industrial standards and guidelines that have been defined by national
and international safety committees, such as IEC/ISA62443 or the IEEE, is recommended.
This includes organizational security measures such as the assignment of accessauthorization as well as technical measures such as network segmentation.

12

Endress+Hauser

Proline Teqwave MW 300 HART

Product description

3

Product description

The device consists of a transmitter and a sensor.
The device is available as a compact version:
The transmitter and sensor form a mechanical unit.

3.1

Product design
2

1

4

3

un
ter

n
f ne
öf

Ni
Spcht Powan un
Do nu te er
en no ngr
er t op öffn
Ne gizeso pa den when
us s outens vr en
ionir

i sp

la y

e

D

Nicht

I/O

ar
ES

C

–
+
E

5
A0052634

1
1
2
3
4
5

Important components of a measuring device
Connection compartment cover
Display module
Transmitter housing
Electronics compartment coverSensorUse of the device with the remote display and operating module DKX001 →  39.

Endress+Hauser

13

Incoming acceptance and product identification

Proline Teqwave MW 300 HART

4

Incoming acceptance and productidentification

4.1

Incoming acceptance
Are the order codes on thedelivery note (1) and the
product sticker (2) identical?

1
2

1
2

Are the goods undamaged?

Does the data on the
nameplate match the orderspecifications on the delivery
note?

Order code:
Ser. no.:Ext. ord. cd.:

i

i

Date:

Is the envelope present withaccompanying documents?

• If one of the conditions is not satisfied, contact your Endress+Hauser Sales Center.
• The Technical Documentation is available via the Internet or via the Endress+Hauser
Operations app: Product identification→  15.

14

Endress+Hauser

Proline Teqwave MW 300 HART

4.2

Incoming acceptance and product identification

Product identification

The following options are available for identification of the device:• Nameplate• Order code with details of the device features on the delivery note
• Enter the serial numbers from the nameplates in the Device Viewer(www.endress.com/deviceviewer): all the information about the device is displayed.
• Enter the serial numbers from the nameplates into the Endress+Hauser Operations app
or scan the DataMatrix code on the nameplate with the Endress+Hauser Operations app:
all the information about the device is displayed.
For an overview of the scope of the associated Technical Documentation, refer to thefollowing:• The "Additional standard documentation on the device" and "Supplementary devicedependent documentation" sections• The Device Viewer: Enter the serial number from the nameplate(www.endress.com/deviceviewer)• The Endress+Hauser Operations app: Enter the serial number from the nameplate orscan the DataMatrix code on the nameplate.

Endress+Hauser

15

Incoming acceptance and product identification

4.2.1

Proline Teqwave MW 300 HART

Transmitter nameplate
1 2

3 4 5

Order code:
Ser. no.:Ext. ord. cd.:

20

6

19
7
18
17
16
15
14

8
i

i

Date:

13

12

11

9

10
A0029192

2
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18
19
20

16

Example of a transmitter nameplate
Manufacturer address/certificate holder
Name of the transmitter
Order code
Serial number (Ser. no.)Extended order code (ext. ord. cd.)
Degree of protection
Space for approvals: use in hazardous areas
Electrical connection data: available inputs and outputs
2-D matrix code
Date of manufacture: year-month
Document number of safety-related supplementary documentation
Space for approvals and certificates: e.g. CE mark, RCM tick
Space for degree of protection of connection and electronics compartment when used in hazardous areas
Firmware version (FW) and device revision (Dev.Rev.) from the factory
Space for additional information in the case of special products
Permitted temperature range for cable
Permitted ambient temperature (Ta)Information on cable gland
Available inputs and outputs, supply voltage
Electrical connection data: supply voltage

Endress+Hauser

Proline Teqwave MW 300 HART

Incoming acceptance and product identification

4.2.2

Sensor nameplate
1

2
3
4
5

Order code:
Ser. no.:Ext. ord. cd.:

6

7
12
11

i

Date:

i

8
9

10
A0051311

1
2
3
4
5
6

7
8
9
10
11
12

Name of sensor
Manufacturer address/certificate holder
Order code
Serial number (Ser. no.)Extended order code (Ext. ord. cd.)
Full scale value; nominal diameter of the sensor; pressure rating; nominal pressure; static pressure; medium
temperature range; measuring tube material, antennas, temperature sensor and gasket between antennaholder and cast body, antenna ceramic
Approval information for explosion protection, Pressure Equipment Directive and degree of protection
Date of manufacture: year-month2-D matrix code
CE mark, RCM-Tick markDocument number of safety-related supplementary documentation
Permitted ambient temperature (Ta)

Order code
The measuring device is reordered using the order code.
Extended order code
• The device type (product root) and basic specifications (mandatory features) arealways listed.
• Of the optional specifications (optional features), only the safety and approvalrelated specifications are listed (e.g. LA). If other optional specifications are alsoordered, these are indicated collectively using the # placeholder symbol (e.g. #LA#).
• If the ordered optional specifications do not include any safety and approval-related
specifications, they are indicated by the + placeholder symbol (e.g. XXXXXX-ABCDE+).

Endress+Hauser

17

Incoming acceptance and product identification

4.2.3
SymbolProline Teqwave MW 300 HART

Symbols on the device
MeaningWARNING!
This symbol alerts you to a dangerous situation. Failure to avoid this situation can result in serious
or fatal injury. Please consult the documentation for the measuring device to discover the type of
potential danger and measures to avoid it.
Reference to documentation
Refers to the corresponding device documentation.
Protective ground connection
A terminal that must be connected to the ground prior to establishing any other connections.

18

Endress+Hauser

Proline Teqwave MW 300 HART

Storage and transport

5

Storage and transport

5.1

Storage conditions

Observe the following notes for storage:‣ Store in the original packaging to ensure protection from shock.
‣ Do not remove protective covers or protective caps installed on process connections.
They prevent mechanical damage to the sealing surfaces and contamination in themeasuring tube.
‣ Protect from direct sunlight to avoid unacceptably high surface temperatures.
‣ Store in a dry and dust-free place.
‣ Do not store outdoors.
Storage temperature →  177

5.2

Transporting the product
Do not remove protective covers or caps installed on process connections. They
prevent mechanical damage to the sealing surfaces and contamination in themeasuring tube.

5.2.1

Measuring devices with lifting lugs

Devices with a nominal diameter of DN 200 to 300 mm (8 to 12 in) have two options for
mounting lifting lugs (eyebolts) for the purpose of transport. The two upper threadedholes are provided for vertically transporting the device, while the two upper threadedholes and one of the opposite lower threaded holes are provided for horizontal transport.

LCAUTION
Special transportation instructions for devices with lifting lugs
‣ Only use the lifting lugs fitted to the device for transport.
‣ The device must always be attached to two lifting lugs when transported vertically andthree lifting lugs when transported horizontally.

A0053150

3Endress+Hauser

Vertical and horizontal transport of the device using mounted lifting lugs

19

Storage and transport

Proline Teqwave MW 300 HART

5.3

Packaging disposal

All packaging materials are environmentally friendly and 100% recyclable:• Outer packaging of device
Stretch wrap made of polymer in accordance with EU Directive 2002/95/EC (RoHS)• Packaging• Wood crate treated in accordance with ISPM 15 standard, confirmed by IPPC logo
• Cardboard box in accordance with European packaging guideline 94/62/EC,
recyclability confirmed by Resy symbol• Transport material and fastening fixtures• Disposable plastic pallet• Plastic straps• Plastic adhesive strips• Filler material
Paper pads

20

Endress+Hauser

Proline Teqwave MW 300 HART

Mounting procedure

6

Mounting procedure

6.1

Mounting requirements

6.1.1

Mounting position

Installation point
Installation in pipe
Do not install the device:• At the highest point of the pipe (risk of gas bubbles accumulating in the measuring tube)• Upstream of a free pipe outlet in a down pipe

A0042131

Install the device:• Ideally in an ascending pipe
• Upstream of an ascending pipe or in areas where the device is filled with medium

A0042317

Installation near valves
Mount the sensor upstream from control valves if possible.

A0041091

Endress+Hauser

21

Mounting procedure

Proline Teqwave MW 300 HART

Installation near pumps• Install the device in the direction of flow downstream from the pump.
• Also install pulsation dampers if reciprocating, diaphragm or peristaltic pumps are used.

A0041083

Installation upstream from a down pipe
If installing upstream of down pipes with a length h ≥ 5 m (16.4 ft): Install a siphon with avent valve downstream of the device.
1h2

A0028981

412hThis arrangement prevents the flow of liquid stopping in the pipe and the formation of air pockets.
Vent valve
Pipe siphon
Length of down pipe

Installation with partially filled pipes• Partially filled pipes with a gradient require a drain-type configuration.
• The installation of a cleaning valve is recommended.

A0047712

Installation in event of pipe vibrations
NOTICE
Pipe vibrations can damage the device!
‣ Do not expose the device to strong vibrations.
Information on the measuring system's resistance to vibration and shock →  177

22

Endress+Hauser

Proline Teqwave MW 300 HART

Mounting procedure

Orientation

1

2
A0052238

1
2

Vertical orientation
Horizontal orientation

Vertical orientation
The device should ideally be installed in a rising pipe:• To avoid having a partially filled pipe
• To avoid any gas accumulation• The measuring tube can be completely drained and protected against the buildup of
deposits.
In the case of total solids of ≥ 20 %TS:
Install the device vertically. If it is installed horizontally, separating layers can form asa result of sedimentation, separating liquid and solids. This can lead to measurement
errors.
Horizontal orientation
The antennas (transmitter and receiver) should be positioned horizontally in order toavoid interference to the measurement signal caused by entrained air bubbles.

3
1

2

A0047713

1
2
3

Antenna - transmitter
Antenna - receiver
Temperature sensor

Flow direction
The device can be installed independently of the flow direction.

Endress+Hauser

23

Mounting procedure

Proline Teqwave MW 300 HART

Inlet and outlet runs
When installing the device, no inlet and outlet runs need to be taken into account. No
special precautions need to be taken for fittings that create turbulence, such as valves,elbows or T-pieces, as long as no cavitation occurs.
Installation with sampling points
To obtain a representative sample, the sampling points should be installed in theimmediate vicinity of the device. This also makes it easier to take the sample and run the
wizards via the device's local operation.

1

1
A0047711

1

Sampling point

Installation with option for cleaning
Depending on the process conditions (e.g. grease deposits), it may be necessary to clean
the device. Additional components can be fitted to avoid any need to remove the device forcleaning:• Rinse connection• Cleaning shaft

1

1

2
A0047740

1
2

Shutoff valve
Shut-off flap for cleaning

If there is a risk of deposits building up in the measuring tube, as a result of grease forexample, a flow velocity of >2 m/s (6.5 ft/s) is recommended.
Dimensions
For the dimensions and installation lengths of the device, see the "Technical
Information" document, "Mechanical construction" section .→  191

24

Endress+Hauser

Proline Teqwave MW 300 HART

Mounting procedure

6.1.2

Environmental and process requirements

Ambient temperature range
Technical data for the ambient temperature range →  176
If operating outdoors:• Install the measuring device in a shady location.
• Avoid direct sunlight, particularly in warm climatic regions.
• Avoid direct exposure to weather conditions.
A weather protection cover is available as an accessory.→  162
Static pressure
Technical data for the static pressure →  178
Vibration and shock resistanceTechnical data for vibration and shock resistance →  177
Thermal insulation• For very hot media: To reduce energy losses and prevent accidental contact with hotpipes• In cold environments: To prevent cooling of the pipe wall and the sensor from theoutside, which could promote the formation of grease deposits

max.

A0052236

LWARNING
Electronics overheating on account of thermal insulation!
‣ Do not insulate the transmitter housing.
‣ Insulation may be provided as far as the connection between the sensor and thetransmitter housing.
‣ Maximum permissible temperature at the lower end of the transmitter housing:
75 °C (167 °F)

Endress+Hauser

25

Mounting procedure

Proline Teqwave MW 300 HART

6.1.3

Special mounting instructions

Weather protection cover
280 (11.0)
134 (5.3)

12 (0.47)

30 (1.18)

48 (1.9)

146 (5.75)

255 (10.0)

A0029553

5

Unit mm (in)A weather protection cover is available as an accessory.→  162

26

Endress+Hauser

Proline Teqwave MW 300 HART

Mounting procedure

6.2

Mounting the measuring device

6.2.1

Preparing the measuring device

1.

Remove all remaining transport packaging.

2.

Remove any protective covers or protective caps present from the sensor.

3.

Remove stick-on label on the electronics compartment cover.

6.2.2

Mounting the sensor

LWARNING
Danger due to improper process sealing!
‣ Ensure that the inside diameters of the gaskets are greater than or equal to that of theprocess connections and piping.
‣ Ensure that the gaskets are clean and undamaged.
‣ Secure the gaskets correctly.
‣ Apply the correct screw tightening torques and comply with the mounting instructions→  28.
Center the sensor between the pipe flanges and mount it in the measuring path.
A mounting kit consisting of screws/mounting bolts, gaskets, nuts and washers canbe ordered as an optional extra:• Directly with the device: Order code for "Accessory enclosed", option PE• Separately as an accessory →  162
1.

Position the device or rotate the transmitter housing so that the cable entries do not
point upwards.
A0029263

2.

Endress+Hauser

Following the correct screw tightening torques and mounting instructions →  28,
mount the sensor between the pipe flanges in the measurement path.

27

Mounting procedure

Proline Teqwave MW 300 HART

4
1
2
3

A0047715

6
1
2
3
4

Mounting the sensor
Nut
WasherScrew/mounting bolt
GasketScrew tightening torques
NOTICE
Failure to observe screw tightening torques or mounting instructions
The process connection may be overloaded if screw tightening torques are not observed or
the mounting instructions cannot be followed. This can produce a leaking processconnection from which the medium escapes.
‣ Apply the correct screw tightening torques and comply with the mounting instructions.
The following mounting instructions must be observed:• The specified screw tightening torques only apply when using the mounting kit, which
can be ordered as an accessory →  163.
• Nuts, threads and screw head surfaces must be greased before assembly.
• The pipes must be free of tensile stress.
• The screws must be tightened evenly in diagonally opposite sequence.
The values for the screw tightening torques depend on variables such as seals, screws,
lubricants, tightening methods etc. These variables are outside the control of the
manufacturer. The values indicated are therefore guideline values only.
Maximum screw tightening torques for EN 1092-1
Nominal diameter

28

[mm]

[in]

50

2

80

3

100

4

150

6

200

8

Pressure rating

ScrewsMax. screw tightening torque

[mm]
PN 10

4 x M1685 Nm (62.7 lbf ft)

8 x M1685 Nm (62.7 lbf ft)

8 x M16100 Nm (73.8 lbf ft)

8 x M20200 Nm (147.5 lbf ft)

PN 10

8 x M20200 Nm (147.5 lbf ft)

PN 16

12 x M20

200 Nm (147.5 lbf ft)

PN 16
PN 10
PN 16
PN 10
PN 16
PN 10
PN 16

Endress+Hauser

Proline Teqwave MW 300 HART

Mounting procedure

Nominal diameter
[mm]

[in]

250

10

300

12

Pressure rating

ScrewsMax. screw tightening torque

[mm]
PN 10

12 x M20

220 Nm (162.3 lbf ft)

PN 16

12 x M24

250 Nm (184.4 lbf ft)

PN 10

12 x M20

220 Nm (162.3 lbf ft)

PN 16

12 x M24

300 Nm (221.3 lbf ft)

Maximum screw tightening torques for ASME B16.5
Nominal diameter

Pressure rating

ScrewsMax. screw tightening torque

[mm]

[in]

[in]

50

2

Class 150

4 x 5/8"

110 Nm (81.1 lbf ft)

80

3

Class 150

4 x 5/8"

130 Nm (95.9 lbf ft)

100

4Class 150

8 x 5/8"

130 Nm (95.9 lbf ft)

150

6Class 150

8 x 3/4"

220 Nm (162.3 lbf ft)

200

8Class 150

8 x 3/4"

250 Nm (184.4 lbf ft)

250

10

Class 150

12 x 7/8"

300 Nm (221.3 lbf ft)

300

12

Class 150

12 x 7/8"

350 Nm (258.2 lbf ft)

ScrewsMax. screw tightening torqueMaximum screw tightening torques for JIS B2220
Nominal diameter

Pressure rating

[mm]

[in]

50

2

10K

4 x M16

90 Nm (66.4 lbf ft)

80

3

10K

8 x M16

90 Nm (66.4 lbf ft)

100

4

10K

8 x M16

90 Nm (66.4 lbf ft)

150

6

10K

8 x M20200 Nm (147.5 lbf ft)

200

8

10K

12 x M20

200 Nm (147.5 lbf ft)

250

10

10K

12 x M22

280 Nm (206.5 lbf ft)

300

12

10K

16 x M22

280 Nm (206.5 lbf ft)

6.2.3

[mm]

Turning the transmitter housing

To provide easier access to the connection compartment or display module, the transmitter
housing can be turned.

Endress+Hauser

29

Mounting procedure

Proline Teqwave MW 300 HART

4.

n
f ne
öf

Nicht

Nicht

un
te r

un
te r

n
f ne
öf
ES
C

ES

C

–

–

+

+

E

e

e

E

ar

ar

3 mm

2.

1.

4 mm 7 Nm (5.2 lbf ft) 3.
A0029993

7

Non-Ex housing1.

Depending on the device version: Loosen the securing clamp of the connection
compartment cover.

2.

Unscrew the connection compartment cover.

3.

Loosen the securing screw.

4.

Turn the housing to the desired position.

5.

Tighten the securing screw.

6.

Screw on the connection compartment cover.

7.

Depending on the device version: Attach the securing clamp of the connection
compartment cover.
2.

A

1.

4 mm 7 Nm (5.2 lbf !)3.

AA0043150

8

Ex housing

1.

Loosen the fixing screws.

2.

Turn the housing to the desired position.

3.

Tighten the securing screws.

6.2.4

Turning the display module

The display module can be turned to optimize display readability and operability.

30

Endress+Hauser

Proline Teqwave MW 300 HART

Mounting procedure

n
f ne
öf

Nicht

Nicht

un
te r

un
te r

n
f ne
öf

ES

ES

C

C

–

–

+

+

ES
C

E

–

E

ar

e

E

ar

2.

3 mm

1.

e

+

3.
A0030035

Endress+Hauser

1.

Depending on the device version: Loosen the securing clamp of the connection
compartment cover.

2.

Unscrew the connection compartment cover.

3.

Turn the display module to the desired position: max. 8 × 45° in each direction.

4.

Screw on the connection compartment cover.

5.

Depending on the device version: Attach the securing clamp of the connection
compartment cover.

31

Mounting procedure

Proline Teqwave MW 300 HART

6.3

Post-mounting check

Is the device undamaged (visual inspection)?



Does the measuring device conform to the measuring point specifications?
For example:• Process temperature• Pressure (refer to the "Pressure-temperature ratings" section of the "Technical Information"
document)• Ambient temperature• Measuring range

Has the correct orientation for the sensor been selected?

32• According to sensor type
• According to medium temperature• According to medium properties

Are the measuring point identification and labeling correct (visual inspection)?

Is the device adequately protected against precipitation and direct sunlight?

Have the fixing screws been tightened with the correct tightening torque?

Endress+Hauser

Proline Teqwave MW 300 HART

7

Electrical connection

Electrical connection

LWARNING
Live parts! Incorrect work performed on the electrical connections can result in anelectric shock.
‣ Set up a disconnecting device (switch or power-circuit breaker) to easily disconnect thedevice from the supply voltage.
‣ In addition to the device fuse, include an overcurrent protection unit with max. 10 A in
the plant installation.

7.1

Electrical safety

In accordance with applicable national regulations.

7.2

Connecting requirements

7.2.1

Required tools• For cable entries: Use corresponding tools• For securing clamp: Allen key 3 mm
• Wire stripper• When using stranded cables: Crimper for wire end ferrule• For removing cables from terminal: Flat blade screwdriver ≤ 3 mm (0.12 in)

7.2.2

Requirements for connection cable

The connecting cables provided by the customer must fulfill the following requirements.
Protective grounding cable for the outer ground terminal
Conductor cross-section < 2.1 mm2 (14 AWG)
The use of a cable lug enables the connection of larger cross-sections.
The grounding impedance must be less than 2 Ω.
Permitted temperature range• The installation guidelines that apply in the country of installation must be observed.
• The cables must be suitable for the minimum and maximum temperatures to be
expected.
Power supply cable (incl. conductor for the inner ground terminal)
Standard installation cable is sufficient.
Signal cable
Current output 4 to 20 mA HARTA shielded cable is recommended. Observe grounding concept of the plant.
Current output 0/4 to 20 mA
Standard installation cable is sufficient
Pulse /frequency /switch output
Standard installation cable is sufficientEndress+Hauser

33

Electrical connection

Proline Teqwave MW 300 HART

Relay output
Standard installation cable is sufficient.
Current input 0/4 to 20 mA
Standard installation cable is sufficient
Status inputStandard installation cable is sufficient
Cable diameter• Cable glands supplied:
M20 × 1.5 with cable ⌀ 6 to 12 mm (0.24 to 0.47 in)• Spring-loaded terminals: Suitable for strands and strands with ferrules.
Conductor cross-section 0.2 to 2.5 mm2 (24 to 12 AWG).
Requirements for connecting cable – remote display and operating module DKX001
Optionally available connecting cable
A cable is supplied depending on the order option• Order code for measuring device: order code 030 for "Display; operation", option Oor• Order code for measuring device: order code 030 for "Display; operation", option M
and• Order code for DKX001: order code 040 for "Cable", option A, B, D, E
Standard cable2 × 2 × 0.34 mm2 (22 AWG) PVC cable with common shield (2 pairs, pairstranded)

Flame resistanceAccording to DIN EN 60332-1-2

Oil resistanceAccording to DIN EN 60811-2-1

ShieldTin-plated copper braid, optical cover ≥ 85 %Capacitance: core/shield≤ 200 pF/m

L/R

≤ 24 µH/Ω

Available cable length

5 m (15 ft)/10 m (35 ft)/20 m (65 ft)/30 m (100 ft)

Operating temperature

When mounted in a fixed position: –50 to +105 °C (–58 to +221 °F); when cable
can move freely: –25 to +105 °C (–13 to +221 °F)

Standard cable - customer-specific cableWith the following order option, no cable is supplied with the device and must be provided
by the customer:
Order code for DKX001: Order code 040 for "Cable", option 1 "None, provided by customer,
max 300 m"
A standard cable with the following minimum requirements can be used as the connecting
cable, even in the hazardous area (Zone 2, Class I, Division 2 and Zone 1, Class I, Division
1):

34Standard cable

4 wires (2 pairs); pair-stranded with common shield, minimum wire cross-section
0.34 mm2 (22 AWG)

ShieldTin-plated copper braid, optical cover ≥ 85 %Cable impedance (pair)

Minimum 80 Ω

Cable length

Maximum 300 m (1 000 ft), maximum loop impedance 20 Ω

Endress+Hauser

Proline Teqwave MW 300 HART

Electrical connection

Capacitance: core/shield

Maximum 1 000 nF for Zone 1, Class I, Division 1

L/R

Maximum 24 µH/Ω for Zone 1, Class I, Division 1

7.2.3

Terminal assignment

Transmitter: supply voltage, input/outputs
The terminal assignment of the inputs and outputs depends on the individual orderversion of the device. The device-specific terminal assignment is documented on anadhesive label in the terminal cover.
Supply voltage
1 (+)

2 (–)

Input/output
1
26 (+)

27 (–)

Input/output
2
24 (+)

25 (–)

Input/output
3
22 (+)

23 (–)

Device-specific terminal assignment: adhesive label in terminal cover.

Terminal assignment of the remote display and operating module →  39.

7.2.4

Preparing the measuring device

NOTICE
Insufficient sealing of the housing!
Operational reliability of the measuring device could be compromised.
‣ Use suitable cable glands corresponding to the degree of protection.
1.

Remove dummy plug if present.

2.

If the measuring device is supplied without cable glands:
Provide suitable cable gland for corresponding connection cable.

3.

If the measuring device is supplied with cable glands:
Observe the requirements for connection cables.

7.3

Connecting the measuring device

NOTICEAn incorrect connection compromises electrical safety!
‣ Only properly trained specialist staff may perform electrical connection work.
‣ Observe applicable federal/national installation codes and regulations.
‣ Comply with local workplace safety regulations.
‣ Always connect the protective ground cable  before connecting additional cables.
‣ When using in potentially explosive atmospheres, observe the information in thedevice-specific Ex documentation.

Endress+Hauser

35

Electrical connection

Proline Teqwave MW 300 HART

7.3.1

Connecting the transmitter
1
2
3
4

A0026781

1
2
3

4

Terminal connection for supply voltage
Terminal connection for signal transmission, input/output
Terminal connection for signal transmission, input/output or terminal connection for network connection via
service interface (CDI-RJ45); optional: connection for external WLAN antenna or remote display andoperating module DKX001
Protective earth (PE)

3.
n
f ne
öf

Nicht

Nicht

un
ter

un
te r

n
f ne
öf
ES
C

ES

C

–

–

+

+

E

E
ES

C

+

ar

ar

4.

1.

3 mm

2.

e

e

–

E

3.
A0029813

1.

Loosen the securing clamp of the connection compartment cover.

2.

Unscrew the connection compartment cover.

3.

Squeeze the tabs of the display module holder together.

4.

Remove the display module holder.

Ni
Spcht Powan un
Do nu te er
en no ngrer töfNe gi op fn
so pa zeden when
us s outens vr en
ionir

un
te r

Nicht

I/O

e

un
te r

6.

er
Powr ente fn nun g öf whe
enichtnnunNpat op
d vrir
S nozeDoergis ouionen patens
Neus
so
I/O

e

Nicht

I/O

n
f ne
öf

n
f ne
öf

Ni
Spcht Powan un
Do nu te er
en no ngrer töfNe gi op fn
so pa zeden when
us s outens vr en
ionir

ar

ar

ES

ES

C

–

C

–
+

+
E

E

5.
A0029814

36

5.

Attach the holder to the edge of the electronics compartment.

6.

Open the terminal cover.

Endress+Hauser

Proline Teqwave MW 300 HART

Electrical connection8.

10 (0.4)

7.

e

Nicht

un
ter

n
f ne
öf

mm (in)

ar
ES
C

–
+

9.

E

A0029815

7.

Push the cable through the cable entry. To ensure tight sealing, do not remove thesealing ring from the cable entry.

8.

Strip the cable and cable ends. In the case of stranded cables, also fit ferrules.

9.

Connect the protective ground.

11.

n
f ne
öf

Nicht

un
te r

22 mm

e

24 mm
ar
ES

C

–
+
E

10.
A0029816

10. Connect the cable according to the terminal assignment.
 Signal cable terminal assignment: The device-specific terminal assignment isdocumented on an adhesive label in the terminal cover.
Supply voltage connection terminal assignment: Adhesive label in theterminal cover or →  35.
11. Firmly tighten the cable glands.
 This concludes the cable connection process.
12. Close the terminal cover.
13. Fit the display module holder in the electronics compartment.
14. Screw on the connection compartment cover.
15. Secure the securing clamp of the connection compartment cover.
Removing a cable
To remove a cable from the terminal:

Endress+Hauser

37

Electrical connection

Proline Teqwave MW 300 HART

3 (0.12)1.

2.

A0029598

938

Engineering unit mm (in)1.

Use a flat-blade screwdriver to press down on the slot between the two terminal
holes.

2.

Remove the cable end from the terminal.

Endress+Hauser

Proline Teqwave MW 300 HART

Electrical connection

7.3.2

Connecting the remote display and operating module DKX001

The remote display and operating module DKX001 is available as an optional extra→  162.

Gnd

Vcc• The measuring device is always supplied with a dummy cover when the remote
display and operating module DKX001 is ordered directly with the measuring
device. Display or operation at the transmitter is not possible in this case.
• If ordered subsequently, the remote display and operating module DKX001 may not
be connected at the same time as the existing measuring device display module.
Only one display or operation unit may be connected to the transmitter at any one
time.

A

B

81 82 83 84
1

3

Gnd

4
5

Vcc

81 82 83 84
A

B

A0027518

1
2
3
4
5

Remote display and operating module DKX001
Terminal connection for potential equalization (PE)Connecting cable
Measuring device
Terminal connection for potential equalization (PE)

7.4

Potential equalization

7.4.1

Requirements

For potential equalization:• Pay attention to in-house grounding concepts• Take account of operating conditions like the pipe material and grounding• Medium, Connect the sensor and transmitter to the same electric potential 1)• Use a ground cable with a minimum cross-section of 6 mm2 (10 AWG) and a cable lug
for potential equalization connections

1)

Endress+Hauser

39

Electrical connection

Proline Teqwave MW 300 HART

7.5

Special connection instructions

7.5.1

Connection examples

Current output 4 to 20 mA HART
1

2

3
4...20 mA
6

4

5
A0029055

 10
1
2
3
4
5
6

Connection example for 4 to 20 mA HART current output (active)

Automation system with current input (e.g. PLC)
Ground cable shield at one end. The cable shield must be grounded at both ends to comply with EMC
requirements; observe cable specifications
Connection for HART operating devices →  66
Resistor for HART communication (≥ 250 Ω): observe maximum load →  169
Analog display unit: observe maximum load →  169
Transmitter

1

2

3

4

5
4...20 mA

A0028762

 11
1
2
3
4
5

40

Connection example for 4 to 20 mA HART current output (passive)

Automation system with current input (e.g. PLC)
Power supply
Ground cable shield at one end. The cable shield must be grounded at both ends to comply with EMC
requirements; observe cable specifications
Analog display unit: observe maximum load →  169
Transmitter

Endress+Hauser

Proline Teqwave MW 300 HART

Electrical connection

HART input
1

2

4

3

6
4...20 mA

5

2

3
A0028763

 12
1
2
3
4
5
6

Connection example for HART input with a common negative (passive)

Automation system with HART output (e.g. PLC)
Active barrier for power supply (e.g. RN221N) →  174
Ground cable shield at one end. The cable shield must be grounded at both ends to comply with EMC
requirements. Observe cable specifications.
Analog display unit: Observe maximum load. →  169
Flowmeter (e.g. Promag W): Observe requirements. →  167
Transmitter

Current output 4-20 mA
1

2

3
4...20 mA
A0028758

 13
1
2
3

Connection example for 4-20 mA current output (active)

Automation system with current input (e.g. PLC)
Analog display unit: observe maximum load →  169
Transmitter

1

2

3

4
4...20 mA

A0028759

 14
1
2
3
4

Endress+Hauser

Connection example for 4-20 mA current output (passive)

Automation system with current input (e.g. PLC)
Active barrier for power supply (e.g. RN221N)
Analog display unit: observe maximum load →  169
Transmitter

41

Electrical connection

Proline Teqwave MW 300 HART

Pulse/frequency output

1

2

3

12345
A0028761

 15
1
2
3

Connection example for pulse/frequency output (passive)

Automation system with pulse/frequency input (e.g. PLC with 10 kΩ pull-up or pull-down resistor)
Power supply
Transmitter: observe input values →  170

Switch output

1

2

3

A0028760

 16
1
2
3

Connection example for switch output (passive)

Automation system with switch input (e.g. PLC with a 10 kΩ pull-up or pull-down resistor)
Power supply
Transmitter: observe input values →  170

Relay output

1

2

3

A0028760

 17
1
2
3

42

Connection example for relay output (passive)

Automation system with relay input (e.g. PLC)
Power supply
Transmitter: observe input values →  171

Endress+Hauser

Proline Teqwave MW 300 HART

Electrical connection

Current input
1

2

3

4

A0028915

 18
1
2
3
4

Connection example for 4 to 20 mA current input

Power supply
Terminal box
External device (for reading in the flow rate value in order to calculate the load rate)
Transmitter

Status input

1

2

3

A0028764

 19
1
2
3

Endress+Hauser

Connection example for status input

Automation system with status output (e.g. PLC)
Power supply
Transmitter

43

Electrical connection

Proline Teqwave MW 300 HART

7.6

Ensuring the degree of protection

The measuring device fulfills all the requirements for the degree of protection IP66/67,
Type 4X enclosure.
To guarantee the degree of protection IP66/67, Type 4X enclosure, carry out the following
steps after electrical connection:
1.

Check that the housing seals are clean and fitted correctly.

2.

Dry, clean or replace the seals if necessary.

3.

Tighten all housing screws and screw covers.

4.

Firmly tighten the cable glands.

5.

To ensure that moisture does not enter the cable entry:
Route the cable so that it loops down before the cable entry ("water trap").
A0029278

6.

44

The cable glands supplied do not ensure housing protection when not in use. They
must therefore be replaced by dummy plus corresponding to the housing protection.

Endress+Hauser

Proline Teqwave MW 300 HART

7.7

Endress+Hauser

Electrical connection

Post-connection check

Are cables or the device undamaged (visual inspection)?



Do the cables used comply with the requirements →  33?

Does the supply voltage match the specifications on the transmitter nameplate →  174?

Is the terminal assignment correct →  35?

Are the power supply and signal cables correctly connected?

Is the protective earthing established correctly?

Is the cable type route completely isolated? Without loops and cross-overs?

Do the cables have adequate strain relief? Are they routed securely?

• Are all cable glands installed, securely tightened and leak-tight?
• Cable run with "water trap" →  44?

Is the sensor connected to the right transmitter?
Check the serial number on the nameplate of the sensor and transmitter.



Are all housing covers installed and securely tightened?

Is the securing clamp securely tightened?

Are dummy plugs inserted in unused cable entries and have transportation plugs been replacedwith dummy plugs?

45

Operation options

Proline Teqwave MW 300 HART

8

Operation options

8.1

Overview of operation options

ESC

-

+

1

E

2

3

4

5

6
A0034513

1
2
3
4
5
6

46

Local operation via display module
Computer with Web browser (e.g. Internet Explorer) or with operating tool (e.g. FieldCare, DeviceCare, AMS
Device Manager, SIMATIC PDM)
Field Xpert SFX350 or SFX370
Field Xpert SMT70
Mobile handheld terminal
Control system (e.g. PLC)

Endress+Hauser

Proline Teqwave MW 300 HART

Operation options

8.2

Structure and function of the operating menu

8.2.1

Structure of the operating menu

For an overview of the operating menu for experts: see the "Description of Device
Parameters" document supplied with the device

Operating menu for operators and maintenances
Language
Operatation!
Language

Operator

Parameter 1
Parameter n
Submenu 1
Submenu n
Setup

Device tag

Wizard n / Parameter nAdvanced setup

Enter access code
Parameter 1

Task-oriented

Maintenance

Wizard 1 / Parameter 1

Parameter n
Submenu 1
Submenu n
Diagnostics

Parameter 1
Parameter n
Submenu 1
Submenu nOperating menu for experts
ExpertAccess status display
Parameter n

Sensor
Input
OutputCommunication

Function-oriented

ExpertSystemApplication
Diagnostics

A0018237-EN

 20

Endress+Hauser

Schematic structure of the operating menu

47

Operation options

Proline Teqwave MW 300 HART

8.2.2

Operating philosophy

The individual parts of the operating menu are assigned to certain user roles (e.g.
operator, maintenance etc.). Each user role contains typical tasks within the device life
cycle.
Menu/parameter
Language

Taskoriented

Operation

Setup

User role and tasks

Content/meaning

Role "Operator", "Maintenance"
Tasks during operation:• Configuration of the operational
display• Reading measured values• Defining the operating language• Defining the Web server operating language• Resetting and controlling totalizers"Maintenance" roleCommissioning:• Configuration of the measurement• Configuration of the inputs andoutputs• Configuration of the communicationinterface

Wizards for fast commissioning:• Configuring the system units• Displaying the I/O configuration• Configuring the inputs• Configuring the outputs• Configuration of the operational display• Configuring the low flow cut off• Configuration of the operational display (e.g. display format, display
contrast)• Resetting and controlling totalizers

Advanced setup• For more customized configuration of the measurement (adaptation to
special measuring conditions)• Configuration of totalizers• Configuration of WLAN settings• Administration (define access code, reset measuring device)
Diagnostics

Expert48

Functionoriented"Maintenance" role
Troubleshooting:• Diagnostics and elimination of process
and device errors• Measured value simulation

Contains all parameters for error detection and analyzing process and device
errors:• Diagnostic list
Contains up to 5 currently pending diagnostic messages.
• Event logbook
Contains event messages that have occurred.
• Device information
Contains information for identifying the device• Measured values
Contains all current measured values.
• Data logging submenu with the "Extended HistoROM" order option
Storage and visualization of measured values• Heartbeat Technology
Verification of device functionality on request and documentation ofverification results• Simulation
Used to simulate measured values or output values.

Tasks that require detailed knowledge ofthe function of the device:• Commissioning measurements underdifficult conditions• Optimal adaptation of themeasurement to difficult conditions• Detailed configuration of thecommunication interface• Error diagnostics in difficult cases

Contains all of the device parameters and allows direct access to these bymeans of an access code. The structure of this menu is based on the function
blocks of the device:• SystemContains all higher-level device parameters that do not affectmeasurement or measured value communication• Sensor
Configuration of the measurement.
• Input
Configuration of the status input• Output
Configuration of the analog current outputs as well as the pulse/frequency
and switch output• Communication
Configuration of the digital communication interface and the Web server• Application
Configuration of the functions that go beyond the actual measurement(e.g. totalizer)• Diagnostics
Error detection and analysis of process and device errors and for device
simulation and Heartbeat Technology.

Endress+Hauser

Proline Teqwave MW 300 HART

Operation options

8.3

Access to operating menu via local display

8.3.1

Operational display
1
2

F

XXXXXXXXX

3

5.53

4

%TS

5
A0051675

1
2
3
4
5

Operational display
Device tag
Status area
Display range for measured values (up to 4 lines)
Operating elements

Status area
The following symbols appear in the status area of the operational display at the top right:• Status signals→  137• F: Failure• C: Function check• S: Out of specification• M: Maintenance required• Diagnostic behavior→  138• : Alarm• : Warning• : Locking (the device is locked via the hardware )• : Communication (communication via remote operation is active)
Display area
In the display area, each measured value is prefaced by certain symbol types for further
description:
Measured variable

Measurement channel
numberDiagnostic behavior

↓

↓

↓

ExampleAppears only if a diagnostics
event is present for thismeasured variable.

Measured variables
SymbolMeaning
Dry matter concentration
Solids load

Endress+Hauser

49

Operation options

Proline Teqwave MW 300 HART

Temperature
Conductivity

The number and display format of the measured variables can be configured via theFormat display parameter (→  97).
Totalizer
SymbolMeaning
Totalizer

OutputSymbolMeaning
Output The measurement channel number indicates which of the outputs is displayed.
Input
SymbolMeaning
Status input

Measurement channel numbers
SymbolMeaning
Measurement channel 1 to 4
The measurement channel number is displayed only if more than one channel is present
for the same measured variable type .

Diagnostic behavior
SymbolMeaning
Alarm• Measurement is interrupted.
• Signal outputs and totalizers assume the defined alarm condition.
• A diagnostic message is generated.
Warning• Measurement is resumed.
• The signal outputs and totalizers are not affected.
• A diagnostic message is generated.

The diagnostic behavior pertains to a diagnostic event that is relevant to the displayedmeasured variable.

50

Endress+Hauser

Proline Teqwave MW 300 HART

Operation options

8.3.2

2

Navigation view
In the submenu

In the wizard

1

1/../Curr. output 1
S
Proc.var.outp
Temperature

2/../Operation
User role
Operator
Locking status
Display

4

4

5

5A0053523-EN

1
2
3
4
5

3

A0053524-EN

Navigation view
Navigation path to current position
Status area
Display area for navigation
Operating elements →  55

Navigation path
The navigation path to the current position is displayed at the top left in the navigation
view and consists of the following elements:• The display symbol for the menu/submenu ( ) or the wizard ( ).
• An omission symbol (/ ../) for operating menu levels in between.
• Name of the current submenu, wizard or parameter
Display symbol

Omission symbol

Parameter

↓

↓

↓

/ ../

Indication

ExampleFor more information about the icons in the menu, refer to the "Display area" section→  51
Status area
The following appears in the status area of the navigation view in the top right corner:• In the submenu
If a diagnostic event is present, the diagnostic behavior and status signal• In the wizard
If a diagnostic event is present, the diagnostic behavior and status signal
For information on the diagnostic behavior and status signal →  137
Display area
Menus
SymbolMeaning
Operation
Is displayed:• In the menu next to the "Operation" selection• At the left in the navigation path in the Operation menu

Endress+Hauser

51

Operation options

Proline Teqwave MW 300 HART

Setup
Is displayed:• In the menu next to the "Setup" selection• At the left in the navigation path in the Setup menu
Diagnosis
Is displayed:• In the menu next to the "Diagnostics" selection• At the left in the navigation path in the Diagnostics menu
ExpertIs displayed:• In the menu next to the "Expert" selection• At the left in the navigation path in the Expert menu

Submenus, wizards, parameters
SymbolMeaning
Submenu
Wizards
Parameters within a wizard

 No display symbol exists for parameters in submenus.
Locking procedure
SymbolMeaningParameter locked
When displayed in front of a parameter name, indicates that the parameter is locked.
• By a user-specific access code
• By the hardware write protection switch

Wizards
SymbolMeaning
Switches to the previous parameter.
Confirms the parameter value and switches to the next parameter.
Opens the editing view of the parameter.

52

Endress+Hauser

Proline Teqwave MW 300 HART

Operation options

8.3.3

Editing view

Numeric editor
1

0
5
+

2

1
6
–

+0.000 Xx
2
3
8
7

4
9

3

4
A0034250

 21
1
2
3
4

For entering values in parameters (e.g. limit values)

Entry display area
Input screen
Confirm, delete or reject entry
Operating elements

Text editor
1

2

XxxxxxxA B C D E F G H I J K
L M N O P Q R S T U V W
X Y Z

6
7

3
5
4

A0034114

 22
1
2
3
4
5
6
7

For entering text in parameters (e.g. device tag)

Entry display area
Current input screen
Change input screen
Operating elements
Move entry position
Delete entry
Reject or confirm entry

Using the operating elements in the editing viewOperating key

Meaning
Minus keyMove the entry position to the left.
Plus keyMove the entry position to the right.

Endress+Hauser

53

Operation options

Proline Teqwave MW 300 HART

Operating key

MeaningEnter key
• Pressing the key briefly confirms your selection.
• Pressing the key for 2 s confirms your entry.
• Pressing the key for > 3 s calls up the wizards: Compare measured value with reference
value.

+

Escape key combination (press keys simultaneously)
Close the editing view without accepting a change.

Input screens
SymbolMeaning
Upper case
Lower caseNumbers
Punctuation marks and special characters: = + – * / ² ³ ¼ ½ ¾ ( ) [ ] < > { }Punctuation marks and special characters: ' " `^. , ; : ? ! % µ ° € $ £ ¥ § @ # / \ I ~ & _Umlauts and accents

Controlling data entries
SymbolMeaningMove entry position
Reject entry
Confirm entry
Delete character immediately to the left of the entry position
Delete character immediately to the right of the entry position
Clear all the characters entered

54

Endress+Hauser

Proline Teqwave MW 300 HART

Operation options

8.3.4

Operating elements

Operating key

Meaning
Minus key
In menu, submenu
Moves the selection bar upwards in a picklist
In wizards
Goes to previous parameterIn the text and numeric editor
Move the entry position to the left.
Plus key
In menu, submenu
Moves the selection bar downwards in a picklist
In wizards
Goes to the next parameterIn the text and numeric editor
Move the entry position to the right.
Enter key
In the operational display• Pressing the key briefly opens the operating menu.
• Pressing the key for > 3 s opens a context menu with the options:• Call up the wizards: Compare measured value with reference value• Activate keypad lockIn menu, submenu• Pressing the key briefly:• Opens the selected menu, submenu or parameter.
• Starts the wizard.
• If help text is open, closes the help text of the parameter.
• Pressing the key for 2 s in a parameter:
If present, opens the help text for the function of the parameter.
In wizards
Opens the editing view of the parameter and confirms the parameter value
In the text and numeric editor• Pressing the key briefly confirms your selection.
• Pressing the key for 2 s confirms your entry.
Escape key combination (press keys simultaneously)
In menu, submenu• Pressing the key briefly:• Exits the current menu level and takes you to the next higher level.
• If help text is open, closes the help text of the parameter.
• Pressing the key for 2 s returns you to the operational display ("home position").

+In wizards
Exits the wizard and takes you to the next higher level
In the text and numeric editor
Exits the Editing view without applying the changes.
Minus/Enter key combination (press and hold down the keys simultaneously)• If keypad lock is active:
Pressing the key for 3 s deactivates the keypad lock.
• If keypad lock is not active:
Pressing the key for 3 s opens the context menu including the option for activating the
keypad lock.

+

8.3.5

Opening the context menu

Using the context menu, the user can call up the following menus quickly and directly fromthe operational display:• Setup• Data backup• Simulation
Endress+Hauser

55

Operation options

Proline Teqwave MW 300 HART

Calling up and closing the context menu
The user is in the operational display.
1.

Press the  and  keys for longer than 3 seconds.
 The context menu opens.
XXXXXXXXXX
Setup
Conf.backup
Simulation
Keylock on
A0034608-EN

2.

Press  +  simultaneously.
 The context menu is closed and the operational display appears.

Calling up the menu via the context menu

56

1.

Open the context menu.

2.

Press  to navigate to the desired menu.

3.

Press  to confirm the selection.
 The selected menu opens.

Endress+Hauser

Proline Teqwave MW 300 HART

Operation options

8.3.6

Navigating and selecting from list

Different operating elements are used to navigate through the operating menu. The
navigation path is displayed on the left in the header. Icons are displayed in front of theindividual menus. These icons are also shown in the header during navigation.
For an explanation of the navigation view with symbols and operating elements→  51
Example: Setting the number of displayed measured values to "2 values"
XXXXXXXXX

20.50
1.

Main menu
Display Language
EnglishOperation
Setup

2.

Main menu
Display Language
Operation
Setup/ ../Operation

3.

User role
Operator
Locking status
Display4.

/ ../Operation
Locking status
Display/ ../Display
Format display
1 value, max.
Contrast display
Display intervall

5.

/ ../Format display
1 value, max.
Bargr. + 1 value
2 values
Val. large+2val.

6.

/ ../Format display
1 value, max.
Bargr. + 1 value
2 values
Val. large+2val.

7.

XXXXXXXXX

8.

2s

10.50
mA
2800
Hz
A0053525-EN

8.3.7

Calling up help text

Help text is available for some parameters and can be called up from the navigation view.
The help text provides a brief explanation of the parameter function and thereby supports
swift and safe commissioning.

Endress+Hauser

57

Operation options

Proline Teqwave MW 300 HART

Calling up and closing the help text
The user is in the navigation view and the selection bar is on a parameter.
1.

Press  for 2 s.
 The help text for the selected parameter opens.
Ent. access code
Enter access code to disable
write protec.

A0014002-EN

 23

2.

Example: Help text for parameter "Enter access code"

Press  +  simultaneously.
 The help text is closed.

8.3.8

Changing the parameters

Parameters can be changed via the numeric editor or text editor.
• Numeric editor: Change values in a parameter, e.g. specifications for limit values.
• Text editor: Enter text in a parameter, e.g. tag name.
A message is displayed if the value entered is outside the permitted value range.
Ent. access code
Invalid or out of range input
value
Min:0
Max:9999
A0014049-EN

For a description of the editing view - consisting of the text editor and numeric editor- with symbols →  53, for a description of the operating elements →  55

8.3.9

User roles and related access authorization

The two user roles "Operator" and "Maintenance" have different write access to theparameters if the customer defines a user-specific access code. This protects the device
configuration via the local display from unauthorized access →  116.
Defining access authorization for user roles
An access code is not yet defined when the device is delivered from the factory. Accessauthorization (read and write access) to the device is not restricted and corresponds to the"Maintenance" user role.

‣ Define the access code.
 The "Operator" user role is redefined in addition to the "Maintenance" user role.
Access authorization differs for the two user roles.
Access authorization to parameters: "Maintenance" user role
Access code status

Read access

Write accessAn access code has not yet been defined(factory setting).



After an access code has been defined.



 1)

1)

58

The user only has write access after entering the access code.

Endress+Hauser

Proline Teqwave MW 300 HART

Operation options

Access authorization to parameters: "Operator" user role
Access code status

Read access

Write access



– 1)

After an access code has been defined.
1)

Despite the defined access code, certain parameters can always be modified and thus are excluded fromthe write protection as they do not affect the measurement: write protection via access code →  116

The user role with which the user is currently logged on is indicated by the Accessstatus parameter. Navigation path: Operation → Access status

8.3.10

Disabling write protection via access code

If the -symbol appears on the local display in front of a parameter, the parameter iswrite-protected by a user-specific access code and its value cannot be changed at the
moment using local operation →  116.
Parameter write protection via local operation can be disabled by entering the user-specificaccess code in the Enter access code parameter (→  101) via the respective access
option.
1.

After you press , the input prompt for the access code appears.

2.

Enter the access code.
 The -symbol in front of the parameters disappears; all previously writeprotected parameters are now re-enabled.

8.3.11

Enabling and disabling the keypad lockThe keypad lock makes it possible to block access to the entire operating menu via local
operation. As a result, it is no longer possible to navigate through the operating menu orchange the values of individual parameters. Users can only read the measured values on
the operational display.
The keypad lock is switched on and off via the context menu.
Switching on the keypad lockThe keypad lock is switched on automatically:• If the device has not been operated via the display for > 1 minute.
• Each time the device is restarted.
To activate the keylock manually:1.

The device is in the measured value display.
Press the  and  keys for 3 seconds.
 A context menu appears.

2.

In the context menu select the Keylock on option.
 The keypad lock is switched on.
If the user attempts to access the operating menu while the keypad lock is active, theKeylock on message appears.

Switching off the keypad lock

‣ The keypad lock is switched on.
Press the  and  keys for 3 seconds.
 The keypad lock is switched off.

Endress+Hauser

59

Operation options

Proline Teqwave MW 300 HART

8.4

Access to operating menu via web browser

8.4.1

Function range

With the integrated web server, the device can be operated and configured via a web
browser service interface (CDI-RJ45) or WLAN interface. The structure of the operating
menu is the same as for the local display. In addition to the measured values, status
information on the device is displayed and can be used to monitor device health.
Furthermore the device data can be managed and the network parameters can be
configured.
A device that has a WLAN interface (can be ordered as an option) is required for the
WLAN connection: order code for "Display; operation", option G "4-line, illuminated; touch
control + WLAN". The device acts as an Access Point and enables communication bycomputer or a mobile handheld terminal.

8.4.2

Requirements

Computer hardware
Hardware

Interface
CDI-RJ45

WLAN

Interface

The computer must have a RJ45
interface. 1)

The operating unit must have a WLAN
interface.

Connection

Standard Ethernet cable

Connection via Wireless LAN.

ScreenRecommended size: ≥12" (depends on the screen resolution)

1)

Recommended cable: CAT5e, CAT6 or CAT7, with shielded plug (e.g. YAMAICHI product; part no. YConProfixPlug63/Prod. ID: 82-006660)

Computer software
Software

Interface
CDI-RJ45

Recommended operating
systems

WLAN• Microsoft Windows 8 or higher.
• Mobile operating systems:• iOS• Android Microsoft Windows XP and Windows 7 is supported.
Web browsers supported• Microsoft Internet Explorer 8 or higher• Microsoft Edge
• Mozilla Firefox• Google Chrome• SafariComputer settings
Settings

Interface
CDI-RJ45

60

WLAN

User rights

Appropriate user rights (e.g. administrator rights) for TCP/IP and proxy server
settings are necessary (e.g. for adjusting the IP address, subnet mask etc.).

Proxy server settings of the
web browser

The web browser setting Use a proxy server for your LAN must be disabled .

Endress+Hauser

Proline Teqwave MW 300 HART

Operation options

Settings

Interface

JavaScript

CDI-RJ45

WLAN

JavaScript must be enabled.

JavaScript must be enabled.

JavaScript cannot be enabled: IfEnter
http://192.168.1.212/servlet/

WLAN display
 The
requires JavaScript

basic.html in the address bar of the webbrowser. A fully functional but simplified
version of the operating menu structure
starts in the web browser.

support.

installing a new firmware version: WhenTo enable correct data display, clear the
temporary memory (cache) underInternet options in the web browser.
Network connections

Only use the active network connections to the measuring device.
Switch off all other network connections suchas WLAN for example.

Switch off all other network
connections.

In the event of connection problems:
Measuring device: Via CDI-RJ45 service interface
Device

CDI-RJ45 service interface

Measuring device

The measuring device has an RJ45 interface.

Web server

Web server must be enabled; factory setting: ON

 For information on enabling the Web server →  65
Measuring device: via WLAN interface
DeviceWLAN interface

Measuring device

The measuring device has a WLAN antenna:• Transmitter with integrated WLAN antenna• Transmitter with external WLAN antenna

Web server

Web server and WLAN must be enabled; factory setting: ON

 For information on enabling the Web server →  65
8.4.3

Connecting the device

Via service interface (CDI-RJ45)
Preparing the measuring device
1.

Depending on the housing version:
Loosen the securing clamp or fixing screw of the housing cover.

2.

Depending on the housing version:
Unscrew or open the housing cover.

3.

Connect the computer to the RJ45 plug via the standard Ethernet cable .

Configuring the Internet protocol of the computer
The following information refers to the default Ethernet settings of the device.
IP address of the device: 192.168.1.212 (factory setting)1.

Endress+Hauser

Switch on the measuring device.

61

Operation options

Proline Teqwave MW 300 HART

2.

Connect the computer to the RJ45 plug via the standard Ethernet cable →  68.

3.

If a 2nd network card is not used, close all the applications on the notebook.
 Applications requiring Internet or a network, such as e-mail, SAP applications,Internet or Windows Explorer.

4.

Close any open Internet browsers.

5.

Configure the properties of the Internet protocol (TCP/IP) as defined in the table:

IP address

192.168.1.XXX; for XXX all numerical sequences except: 0, 212 and 255 → e.g.
192.168.1.213

Subnet mask

255.255.255.0

Default gateway

192.168.1.212 or leave cells empty

Via WLAN interface
Configuring the Internet protocol of the mobile terminal
NOTICE
If the WLAN connection is lost during the configuration, settings made may be lost.
‣ Make sure that the WLAN connection is not disconnected while configuring the device.
NOTICE
Note the following to avoid a network conflict:‣ Avoid accessing the measuring device simultaneously from the same mobile terminal
via the service interface (CDI-RJ45) and the WLAN interface.
‣ Only activate one service interface (CDI-RJ45 or WLAN interface).
‣ If simultaneous communication is necessary: configure different IP address ranges, e.g.
192.168.0.1 (WLAN interface) and 192.168.1.212 (CDI-RJ45 service interface).
Preparing the mobile terminal‣ Enable WLAN on the mobile terminal.
Establishing a WLAN connection from the mobile terminal to the measuring device
1.

In the WLAN settings of the mobile terminal:
Select the measuring device using the SSID (e.g. EH__300_A802000).

2.

If necessary, select the WPA2 encryption method.

3.

Enter the password:
Serial number of the measuring device ex-works (e.g. L100A802000).
 The LED on the display module flashes. It is now possible to operate themeasuring device with the web browser, FieldCare or DeviceCare.
The serial number can be found on the nameplate.
To ensure the safe and swift assignment of the WLAN network to the measuring
point, it is advisable to change the SSID name. It should be possible to clearly assign
the new SSID name to the measuring point (e.g. tag name) because it is displayed as
the WLAN network.

Terminating the WLAN connection‣ After configuring the device:
Terminate the WLAN connection between the mobile terminal and measuring device.
Starting the web browser
1.

62

Start the web browser on the computer.

Endress+Hauser

Proline Teqwave MW 300 HART

2.

Operation options

Enter the IP address of the web server in the address line of the web browser:
192.168.1.212
 The login page appears.
2 3 4

1

5

Device name:
Device tag:
Signal Status:

Web server languageiEnglish6

Maintenance

7

Login
Access Status
Enter access code

8

i
Login

Reset access code

910A0053670

1
2
3
4
5
6
7
8
9
10

Picture of device
Device name
Device tag (→  82)
Status signal
Current measured values
Operating language
User role
Access code
Login
Reset access code (→  112)

If a login page does not appear, or if the page is incomplete

8.4.4

Logging on

1.

Select the preferred operating language for the Web browser.

2.

Enter the user-specific access code.

3.

Press OK to confirm your entry.

Access code

0000 (factory setting); can be changed by customer

If no action is performed for 10 minutes, the Web browser automatically returns to
the login page.

Endress+Hauser

63

Operation options

Proline Teqwave MW 300 HART

8.4.5

User interface
1

2 3 4

5

6

Device name:
Device tag:
Signal Status:
Total solids adjustment Measured values Menu

Instrument health status Data management

Network

Main Menü
Display languagei

Operation

7

EnglishSetup

Diagnostic

8

Expert

A0053669

1
2
3
4
5
6
7
8

Picture of device
Device name
Device tag
Status signal
Current measured values
Function row
Local display language
Navigation area

Header
The following information appears in the header:• Device name• Device tag• Device status with status signal →  140• Current measured values
Function row
Functions

Meaning

Adjustment ofsolids content

Call up the wizards: adjust measured value based on reference value

Measured values

Displays the measured values of the device

Menu

Device status64• Access to the operating menu from the measuring device• The structure of the operating menu is the same as for the local display
information on the operating menu structure: Description of Device
 Detailed
Parameters
Displays the diagnostic messages currently pending, listed in order of priority

Endress+Hauser

Proline Teqwave MW 300 HART

Operation options

Functions

Meaning

Data exchange between computer and measuring device:• Device configuration:• Load settings from the device(XML format, save configuration)• Save settings to the device(XML format, restore configuration)
Data management • Logbook - Export Event logbook (.csv file)• Documents - Export documents:• Export backup data record(.csv file, create documentation of the measuring point configuration)• Verification report(PDF file, only available with the "Heartbeat Verification" application package)• Firmware update - Flashing a firmware version

Network

Configuration and checking of all the parameters required for establishing the connection
to the measuring device:• Network settings (e.g. IP address, MAC address)• Device information (e.g. serial number, firmware version)

Logout

End the operation and call up the login page

Navigation area
The menus, the associated submenus and parameters can be selected in the navigationarea.
Working area
Depending on the selected function and the related submenus, various actions can be
performed in this area:• Configuring parameters• Reading measured values• Calling up help text
• Starting an upload/download

8.4.6

Disabling the Web server

The Web server of the measuring device can be switched on and off as required using the
Web server functionality parameter.
Navigation"Expert" menu → Communication → Web server
Parameter overview with brief description
Parameter
Web server functionality

Endress+Hauser

Description
Switch the Web server on and off.

Selection• Off• HTML Off• On

Factory settingOn

65

Operation options

Proline Teqwave MW 300 HART

Function scope of the "Web server functionality" parameter
OptionDescription

Off• The Web server is completely disabled.
• Port 80 is locked.

HTML OffThe HTML version of the Web server is not available.

On• The complete Web server functionality is available.
• JavaScript is used.
• The password is transferred in an encrypted state.
• Any change to the password is also transferred in an encrypted state.

Enabling the Web server
If the Web server is disabled it can only be re-enabled with the Web server functionality
parameter via the following operating options:• Via local display• Via Bedientool "FieldCare"• Via "DeviceCare" operating tool

8.4.7

Logging out

Before logging out, perform a data backup via the Data management function(upload configuration from device) if necessary.
1.

Select the Logout entry in the function row.
 The home page with the Login box appears.

2.

Close the Web browser.

3.

If no longer needed:
Reset the modified properties of the Internet protocol (TCP/IP) →  61.

8.5

Access to the operating menu via the operating tool

The structure of the operating menu in the operating tools is the same as for operation via
the local display.

8.5.1

Connecting the operating tool

Via HART protocol
This communication interface is available in device versions with a HART output.

66

Endress+Hauser

Proline Teqwave MW 300 HART

Operation options

8
4

1

2

7

3

5

6
A0028747

 24
1
2
3

4
5
6
7
8

Options for remote operation via HART protocol (active)

Control system (e.g. PLC)
Field Communicator 475
Computer with web browser (e.g. Microsoft Edge) to access the integrated device web server or computer with
operating tool (e.g. FieldCare, DeviceCare, AMS Device Manager, SIMATIC PDM) with COM DTM "CDI
Communication TCP/IP"
Commubox FXA195 (USB)
Field Xpert SFX350 or SFX370
Field Xpert SMT70
VIATOR Bluetooth modem with connecting cable
Transmitter

10
3

2

1

6

4

9

5

7

8
A0028746

 25
1
2
3
4
5

6
7
8
9
10

Endress+Hauser

Options for remote operation via HART protocol (passive)

Control system (e.g. PLC)
Transmitter power supply unit, e.g. RN221N (with communication resistor)
Connection for Commubox FXA195 and Field Communicator 475
Field Communicator 475
Computer with web browser (e.g. Microsoft Edge) to access the integrated device web server or computer with
operating tool (e.g. FieldCare, DeviceCare, AMS Device Manager, SIMATIC PDM) with COM DTM "CDI
Communication TCP/IP"
Commubox FXA195 (USB)
Field Xpert SFX350 or SFX370
Field Xpert SMT70
VIATOR Bluetooth modem with connecting cable
Transmitter

67

Operation options

Proline Teqwave MW 300 HART

Service interface
Via service interface (CDI-RJ45)
A point-to-point connection can be established to configure the device onsite. With the
housing open, the connection is established directly via the service interface (CDI-RJ45) of
the device.
An adapter for the RJ45 to the M12 plug is optionally available for the non-hazardous
area:
Order code for "Accessories", option NB: "Adapter RJ45 M12 (service interface)"
The adapter connects the service interface (CDI-RJ45) to an M12 plug mounted in thecable entry. The connection to the service interface can be established via an M12
plug without opening the device.

1
3

2

A0027563

 26
1
2
3

Connection via service interface (CDI-RJ45)

Computer with web browser (e.g. Microsoft Internet Explorer, Microsoft Edge) for accessing the integrated
web server or with "FieldCare" operating tool, "DeviceCare" with COM DTM "CDI Communication TCP/IP"
Standard Ethernet connecting cable with RJ45 plug
Service interface (CDI-RJ45) of the measuring device with access to the integrated Web server

Via WLAN interface
The optional WLAN interface is available on the following device version:
Order code for "Display; operation", option G "4-line, illuminated; touch control + WLAN"

68

Endress+Hauser

Proline Teqwave MW 300 HART

Operation options

3

1

4

5

2

6

3

4

7
A0034570

1
2
3
4
5
6
7

Transmitter with integrated WLAN antenna
Transmitter with external WLAN antenna
LED lit constantly: WLAN reception is enabled on measuring device
LED flashing: WLAN connection established between operating unit and measuring device
Computer with WLAN interface and web browser (e.g. Microsoft Internet Explorer, Microsoft Edge) foraccessing the integrated device web server or with operating tool (e.g. FieldCare, DeviceCare)
Mobile handheld terminal with WLAN interface and web browser (e.g. Microsoft Internet Explorer, Microsoft
Edge) for accessing the integrated device web server or operating tool (e.g. FieldCare, DeviceCare)
Smart phone or tablet (e.g. Field Xpert SMT70)

Function

WLAN: IEEE 802.11 b/g (2.4 GHz)

Encryption

WPA2-PSK AES-128 (in accordance with IEEE 802.11i)

Configurable WLAN channels

1 to 11Degree of protection

IP67

Available antennas• Internal antenna• External antenna (optional)
In the event of poor transmission/reception conditions at the place ofinstallation.

 Only 1 antenna is active at any one time!
Range• Internal antenna: typically 10 m (32 ft)• External antenna: typically 50 m (164 ft)

Materials (external antenna)• Antenna: ASA plastic (acrylonitrile styrene acrylate) and nickel-plated
brass• Adapter: Stainless steel and nickel-plated brass• Cable: Polyethylene• Plug: Nickel-plated brass• Angle bracket: Stainless steel

Configuring the Internet protocol of the mobile terminal
NOTICE
If the WLAN connection is lost during the configuration, settings made may be lost.
‣ Make sure that the WLAN connection is not disconnected while configuring the device.
NOTICE
Note the following to avoid a network conflict:‣ Avoid accessing the measuring device simultaneously from the same mobile terminal
via the service interface (CDI-RJ45) and the WLAN interface.
‣ Only activate one service interface (CDI-RJ45 or WLAN interface).
‣ If simultaneous communication is necessary: configure different IP address ranges, e.g.
192.168.0.1 (WLAN interface) and 192.168.1.212 (CDI-RJ45 service interface).

Endress+Hauser

69

Operation options

Proline Teqwave MW 300 HART

Preparing the mobile terminal‣ Enable WLAN on the mobile terminal.
Establishing a WLAN connection from the mobile terminal to the measuring device
1.

In the WLAN settings of the mobile terminal:
Select the measuring device using the SSID (e.g. EH__300_A802000).

2.

If necessary, select the WPA2 encryption method.

3.

Enter the password:
Serial number of the measuring device ex-works (e.g. L100A802000).
 The LED on the display module flashes. It is now possible to operate themeasuring device with the web browser, FieldCare or DeviceCare.
The serial number can be found on the nameplate.
To ensure the safe and swift assignment of the WLAN network to the measuring
point, it is advisable to change the SSID name. It should be possible to clearly assign
the new SSID name to the measuring point (e.g. tag name) because it is displayed as
the WLAN network.

Terminating the WLAN connection‣ After configuring the device:
Terminate the WLAN connection between the mobile terminal and measuring device.

8.5.2

Field Xpert SFX350, SFX370

Function scope
Field Xpert SFX350 and Field Xpert SFX370 are mobile computers for commissioning andmaintenance. They enable efficient device configuration and diagnostics for HART and
FOUNDATION Fieldbus devices in the non-hazardous area (SFX350, SFX370) andhazardous area (SFX370).
For details, see Operating Instructions BA01202S
Source for device description files
See information →  74

8.5.3

FieldCare

Function range
FDT-based (Field Device Technology) plant asset management tool from Endress+Hauser.
It can configure all smart field units in a system and helps you manage them. By using thestatus information, it is also a simple but effective way of checking their status andcondition.
Access is via:
• HART protocol →  66
• CDI-RJ45 service interface →  68
• WLAN interface →  68

70

Endress+Hauser

Proline Teqwave MW 300 HART

Operation options

Typical functions:• Transmitter parameter configuration• Loading and saving of device data (upload/download)• Documentation of the measuring point• Visualization of the measured value memory (line recorder) and event logbook• Operating Instructions BA00027S• Operating Instructions BA00059S
Source for device description files →  74
Establishing a connection
1.

Start FieldCare and launch the project.

2.

In the network: Add a device.
 The Add device window opens.

3.

Select the CDI Communication TCP/IP option from the list and press OK to confirm.

4.

Right-click CDI Communication TCP/IP and select the Add device option in thecontext menu that opens.

5.

Select the desired device from the list and press OK to confirm.
 The CDI Communication TCP/IP (Configuration) window opens.

6.

Enter the device address in the IP address field: 192.168.1.212 and press Enter toconfirm.

7.

Establish the online connection to the device.
• Operating Instructions BA00027S• Operating Instructions BA00059S

User interface

Device tag
xxxxxxTotal solid value
Status signal
Maintenance required (M)0.80 %TS

Device namexxxxxx

Locking status
Unlocked

Output current
12.00 mA

> Setup
Setup

Device unit

System units
Current output 1
I/O configuration
Commissioning
Total solids adjustment
Advanced setup%TS

Density unit
g/l

Temperature unit
°C

Conductivity unit
μS/cm?
Device unit%TS
ppm TS
g/l
mg/l
kg/m³
mg/m³
lb/gal (us)lb/ft³

A0053667

8.5.4

DeviceCare

Function range
Tool for connecting and configuring Endress+Hauser field devices.

Endress+Hauser

71

Operation options

Proline Teqwave MW 300 HART

The fastest way to configure Endress+Hauser field devices is with the dedicated"DeviceCare" tool. Together with the device type managers (DTMs) it presents a convenient,comprehensive solution.
Innovation brochure IN01047S
Source for device description files →  74

8.5.5

AMS Device Manager

Function range
Program from Emerson Process Management for operating and configuring measuring
devices via the HART protocol.
Source for device description files →  74

8.5.6

Field Communicator 475

Function scope
Industrial handheld terminal from Emerson Process Management for remoteconfiguration and measured value display via HART protocol.
Source for device description files
See information →  74

8.5.7

SIMATIC PDM

Function range
SIMATIC PDM is a standardized, manufacturer-independent program from Siemens forthe operation, configuration, maintenance and diagnosis of intelligent field devices via
HART protocol.
Source for device description files →  74

72

Endress+Hauser

Proline Teqwave MW 300 HART

System integration

9

System integration

9.1

Overview of device description files

9.1.1

Current version data for the device

Firmware version

01.00.zz• On the title page of the manual• On the transmitter nameplate• Firmware version
Diagnostics → Device information → Firmware
versionRelease date of firmware version

03.2024---

Manufacturer ID

0x11

Manufacturer ID
Diagnostics → Device information → Manufacturer ID

Device type code

11B3

Device type
Diagnostics → Device information → Device type

HART protocol revision

7

HART revision
Expert → Communication → HART output→ Information → HART revision

Device revision1• On the transmitter nameplate• Device revision
Diagnostics → Device information → Device revision

For an overview of the various firmware versions for the device

Endress+Hauser

73

System integration

Proline Teqwave MW 300 HART

9.1.2

Operating tools

Different operating tools can be used for local or remote access to the measuring device.
Depending on the operating tool used, access is possible with different operating units andvia a variety of interfaces.
Supported operating
tools

Operating unit

Interface

Additional information

Web browser

Notebook, PC or tablet
with web browser• CDI-RJ45 service
interface• WLAN interface

Special Documentation for thedevice →  192

DeviceCare SFE100

Notebook, PC or tablet• CDI-RJ45 service
with Microsoft Windowsinterfacesystem• WLAN interface• Fieldbus protocol

Service-specific accessories→  164

Notebook, PC or tablet• CDI-RJ45 service
with Microsoft Windowsinterfacesystem• WLAN interface• Fieldbus protocol

Service-specific accessories→  164

FieldCare SFE500

Sources for obtaining device
descriptions
www.endress.com →Download-Area

Sources for obtaining device
descriptions
www.endress.com →Download-Area

Other operating tools based on FDT technology with a device driver such as DTM/
iDTM or DD/EDD can be used for device operation. These operating tools are available
from the individual manufacturers. Integration into the following operating tools,
among others, is supported:• FactoryTalk AssetCentre (FTAC) from Rockwell Automation →www.rockwellautomation.com• Process Device Manager (PDM) from Siemens → www.siemens.com• Asset Management Solutions (AMS) from Emerson → www.emersonprocess.com• FieldCommunicator 375/475 from Emerson → www.emersonprocess.com• Field Device Manager (FDM) from Honeywell → www.process.honeywell.com• FieldMate from Yokogawa → www.yokogawa.com• PACTWare → www.pactware.com
The related device description files are available: www.endress.com → Download Area

74

Endress+Hauser

Proline Teqwave MW 300 HART

System integration

9.2

Measured variables via HART protocol

9.2.1

Dynamic variables

Measured variables can be assigned to the dynamic variables (PV, SV, TV and QV). This
assignment of variables can take place via local operation or via an operating tool.
The following parameters are available for the assignment of variables:• Assign PV parameter (primary dynamic variable)• Assign SV parameter (second dynamic variable)• Assign TV parameter (third dynamic variable)• Assign QV parameter (fourth dynamic variable)
Navigation"Expert" menu → Communication → HART output → Output
Parameter overview with brief description
Parameter

Prerequisite

Description

Selection

Factory setting

Assign PV

The Load rate option is only
available if the volume flow of
the medium is read in via theCurrent input 1 to n or thefieldbus.

Assign measured variable to• Offprimary dynamic variable (PV). • Total solids• Temperature• Electronics
temperature• Conductivity• Corrected
conductivity• Load rate *Total solids

Assign SV

The Load rate option is only
available if the volume flow of
the medium is read in via theCurrent input 1 to n or thefieldbus.

Assign measured variable tothe second dynamic variable(SV).

Temperature

Endress+Hauser• Conductivity• Corrected
conductivity• Temperature• Electronics
temperature• Total solids• Load rate *• Volume flow *• Totalizer 1 *• Current input 1 *• Current input 2 *• Current input 3 *• HART input

75

System integration

Proline Teqwave MW 300 HART

Parameter

Prerequisite

Description

Selection

Assign TV

The Load rate option is only
available if the volume flow of
the medium is read in via theCurrent input 1 to n or thefieldbus.

Assign measured variable tothe third dynamic variable(TV).

• Conductivity• Corrected
conductivity• Temperature• Electronics
temperature• Total solids• Load rate *• Volume flow *• Totalizer 1 *• Current input 1 *• Current input 2 *• Current input 3 *• HART input

Electronics
temperature

Assign QV

The Load rate option is only
available if the volume flow of
the medium is read in via theCurrent input 1 to n or thefieldbus.

Assign measured variable toquaternary (fourth) dynamic
variable (QV).

• Conductivity• Corrected
conductivity• Temperature• Electronics
temperature• Total solids• Load rate *• Volume flow *• Totalizer 1 *• Current input 1 *• Current input 2 *• Current input 3 *• HART input

Conductivity*

Factory setting

Visibility depends on order options or device settings

9.2.2

Device variables

Measured variables are permanently assigned to the device variables. This assignment
cannot be changed.
The following measured variables are assigned to the device variables:• 0 = conductivity• 1 = corrected conductivity• 2 = temperature• 3 = electronic temperature• 4 = total solids• 5 = load rate
• 6 = volume flow
• 7 = totalizer• 8 = current input 1• 9 = current input 2• 10 = current input 3• 11 = catch device variable• 12 = percent range• 13 = current output
Only a maximum of 8 device variables can be transmitted.

76

Endress+Hauser

Proline Teqwave MW 300 HART

System integration

9.3

Other settings

Burst mode functionality in accordance with HART 7 Specification:
Navigation"Expert" menu → Communication → HART output → Burst configuration → Burstconfiguration 1 to n

‣ Burst configuration 1 to n
Burst mode 1 to n

→  77

Burst command 1 to n

→  77

Burst variable 0→  78Burst variable 1→  78Burst variable 2→  78Burst variable 3→  78Burst variable 4→  78Burst variable 5→  78Burst variable 6→  78Burst variable 7→  78Burst trigger mode→  78Burst trigger level→  78Min. update period→  79Max. update period→  79Parameter overview with brief description
Parameter

Prerequisite

Description

Selection / User
entry

Factory setting

Burst mode 1 to n–

Activate the HART burst modefor burst message X.

• Off
• On

Off

Burst command 1 to n–

Select the HART commandthat is sent to the HART
master.

• Command 1• Command 2• Command 3• Command 9• Command 33• Command 48

Command 2

Endress+Hauser

77

System integration

Proline Teqwave MW 300 HART

Parameter

Prerequisite

Description

Selection / User
entry

Burst variable 0

The Load rate option is only
available if the volume flow of
the medium is read in via theCurrent input 1 to n or thefieldbus.

For HART command 9 and 33:
select the HART device
variable or the process
variable.

• Conductivity• Corrected
conductivity• Temperature• Electronics
temperature• Total solids• Load rate *• Totalizer 1• Percent of range• Measured current• Current input 1 *• Current input 2 *• Current input 3 *• Primary variable(PV)• Secondary variable(SV)• Tertiary variable(TV)• Quaternary
variable (QV)
• HART input• Not used

Total solids

Burst variable 1–

For HART command 9 and 33:
select the HART device
variable or the process
variable.

See the Burst
variable 0
parameter.

Not used

Burst variable 2–

For HART command 9 and 33:
select the HART device
variable or the process
variable.

See the Burst
variable 0
parameter.

Not used

Burst variable 3–

For HART command 9 and 33:
select the HART device
variable or the process
variable.

See the Burst
variable 0
parameter.

Not used

Burst variable 4–

For HART command 9: selectthe HART device variable or
the process variable.

See the Burst
variable 0
parameter.

Not used

Burst variable 5–

For HART command 9: selectthe HART device variable or
the process variable.

See the Burst
variable 0
parameter.

Not used

Burst variable 6–

For HART command 9: selectthe HART device variable or
the process variable.

See the Burst
variable 0
parameter.

Not used

Burst variable 7–

For HART command 9: selectthe HART device variable or
the process variable.

See the Burst
variable 0
parameter.

Not used

Burst trigger mode–

Select the event that triggers
burst message X.

• Continuous• Window *• Rising *• Falling *• On change

Continuous

Burst trigger level–

Enter the burst trigger value.

Signed floating-point
number–Together with the option
selected in the Burst trigger
mode parameter the burst
trigger value determines thetime of burst message X.

78

Factory setting

Endress+Hauser

Proline Teqwave MW 300 HART

Parameter

System integration

Prerequisite

Description

Selection / User
entry

Factory setting

Min. update period–

Enter the minimum time span
between two burst commandsof burst message X.

Positive integer

1 000 ms

Max. update period–

Enter the maximum time span
between two burst commandsof burst message X.

Positive integer

2 000 ms*

Visibility depends on order options or device settings

Endress+Hauser

79Commissioning

Proline Teqwave MW 300 HART

10

Commissioning

10.1

Post-mounting and post-connection check

Before commissioning the device:‣ Make sure that the post-installation and post-connection checks have been performed
successfully.

• Checklist for "Post-mounting" check→  32
• Checklist for "Post-connection check" →  45

10.2

Switching on the measuring device‣ Switch on the device upon successful completion of the post-mounting and post-connection check.
 After a successful startup, the local display switches automatically from the startup
display to the operational display.
If nothing appears on the local display or if a diagnostic message is displayed, refer tothe section on "Diagnostics and troubleshooting" .

10.3

Connecting via FieldCare• For connecting FieldCare →  68
• For connecting via FieldCare →  71
• For user interface of FieldCare →  71

10.4

Setting the operating language

Factory setting: English or ordered local language
XXXXXXXXX

20.50
XX

1.

XX
Main menu
Display language
EnglishOperation
Setup

2.

Display language
Ã English
DeutschEspañol
Français

3.

4.

Display language
Ã English
DeutschEspañol
Français
Hauptmenü
SpracheDeutschBetrieb
Setup
A0053789

 27

80

Using the example of the local display

Endress+Hauser

Proline Teqwave MW 300 HART

Commissioning

10.5

Configuring the measuring device

The Setup menu with its guided wizards contains all the parameters needed for standard
operation.
XXXXXXXXX

20.50
mA
Main menu
Display language

1.

EnglishDisplay/operat.
Setup
Main menu
Display/operat.
Setup

2.

Diagnostic/ ../Setup
Medium selection

3.

XXXXXXXXXXXXXXXXXX
A0032222-EN

 28

Navigation to "Setup" menu using the example of the local display

The number of submenus and parameters can vary depending on the device version.
Certain submenus and parameters in these submenus are not described in the
Operating Instructions. Instead a description is provided in the Special Documentation
for the device ("Supplementary documentation").

 Setup
Device tag→  82‣ System units→  83‣ I/O configuration→  84‣ Current input 1 to n

→  85‣ Status input 1 to n

→  86‣ Current output 1 to n

→  87‣ Pulse/frequency/switch output→  89‣ Relay output 1 to n

→  95‣ Display→  96‣ Total solids commissioning→  991 to nEndress+Hauser

81Commissioning

Proline Teqwave MW 300 HART

10.5.1‣ Total solids adjustment→  99‣ Advanced setup→  100

Defining the tag nameTo enable fast identification of the measuring point within the system, you can enter aunique designation using the Device tag parameter and thus change the factory setting.
1

XXXXXXXXX

A0029422

 29
1

Header of the operational display with tag nameTag nameEnter the tag name in the "FieldCare" operating tool →  71
Navigation"Setup" menu → Device tag
Parameter overview with brief description
Parameter
Device tag

82

Description
Enter the name for the measuring point.

User entry
Max. 32 characters, such as
letters, numbers or special
characters (e.g. @, %, /).

Factory setting
Teqwave M

Endress+Hauser

Proline Teqwave MW 300 HART

Commissioning

10.5.2

Setting the system units

In the System units submenu the units of all the measured values can be set.
The number of submenus and parameters can vary depending on the device version.
Certain submenus and parameters in these submenus are not described in the
Operating Instructions. Instead a description is provided in the Special Documentation
for the device ("Supplementary documentation").
Navigation"Expert" menu → Sensor → System units‣ System units
Total solids unit

→  83

Density unit

→  83Mass flow unit

→  83Mass unit→  83

Volume flow unit

→  83Temperature unit

→  83

Conductivity unit

→  83Date/time format→  83

Parameter overview with brief description
Parameter

Prerequisite

Description

Selection

Factory setting

Total solids unit–

Select total solids unit.

Unit choose list

Depends on country

Density unit–

Select density unit.

Unit choose list

Depends on country

Mass flow unit

The volume flow of themedium is read in via theCurrent input 1 to n.

Select mass flow unit.

Unit choose list

Depends on country

Mass unit

The volume flow of themedium is read in via theCurrent input 1 to n or thefieldbus.

Select mass unit.

Unit choose list

Depends on country

Volume flow unit

The volume flow of themedium is read in via theCurrent input 1 to n.

Select volume flow unit.

Unit choose list

l/hTemperature unit–

Select temperature unit.

Unit choose list

Depends on country

Conductivity unit–

Select conductivity unit.

Unit choose list

µS/cm

Date/time format–

Select date and time format.

• dd.mm.yy hh:mm• dd.mm.yy hh:mm
am/pm• mm/dd/yy hh:mm• mm/dd/yy hh:mm
am/pm

dd.mm.yy hh:mm

Endress+Hauser

83Commissioning

Proline Teqwave MW 300 HART

10.5.3

Displaying the I/O configuration

The I/O configuration submenu guides the user systematically through all the parametersin which the configuration of the I/O modules is displayed.
Navigation"Setup" menu → I/O configuration‣ I/O configuration
I/O module 1 to n terminal numbers→  84I/O module 1 to n information→  84I/O module 1 to n type→  84

Apply I/O configuration→  84I/O alteration code→  84

Parameter overview with brief description
Parameter

Description

User interface / Selection /User entry

Factory setting

I/O module 1 to n terminal numbers Shows the terminal numbers used by the I/O • Not used
module.
• 26-27 (I/O 1)• 24-25 (I/O 2)• 22-23 (I/O 3)–

I/O module 1 to n information

Shows information of the plugged I/O
module.

• Not plugged• Invalid• Not configurable• Configurable• HART–

I/O module 1 to n typeShows the I/O module type.

• Off• Current output *• Current input *• Status input *• Pulse/frequency/switch
output *• Relay output *Off

Apply I/O configuration

Apply parameterization of the freely
configurable I/O module.

• No
• Yes

No

I/O alteration code

Enter the code in order to change the I/O
configuration.

Positive integer

0

*

84

Visibility depends on order options or device settings

Endress+Hauser

Proline Teqwave MW 300 HART

Commissioning

10.5.4

Configuring the current input

The "Current input" wizard guides the user systematically through all the parameters thathave to be set for configuring the current input.
Navigation"Setup" menu → Current input‣ Current input 1 to n
Terminal number→  85Signal mode→  850/4 mA value→  8520 mA value→  85Current span

→  85

Failure mode→  85

Failure value→  85

Parameter overview with brief description
Parameter

Prerequisite

Description

User interface /Selection / User
entry

Factory setting

Terminal number–

Shows the terminal numbers
used by the current input
module.

• Not used
• 24-25 (I/O 2)• 22-23 (I/O 3)–

Signal mode–

Select the signal mode for thecurrent input.

• Passive• Active *Passive

0/4 mA value–Enter 4 mA value.

Signed floating-point
number

0 % TS

20 mA value–Enter 20 mA value.

Signed floating-point
number

12 %TS

Current span–

Select current range forprocess value output andupper/lower level for alarm
signal.

• 4...20 mA (4...20.5 Country-specific:mA)• 4...20 mA NE• 4...20 mA NE(3.8...20.5 mA)(3.8...20.5 mA)• 4...20 mA US• 4...20 mA US(3.9...20.8 mA)(3.9...20.8 mA)• 0...20 mA (0...20.5
mA)

Failure mode–

Define input behavior in alarm
condition.

• Alarm• Last valid value• Defined value

Alarm

Failure value

In the Failure modeparameter, the Defined value
option is selected.

Enter value to be used by thedevice if input value fromexternal device is missing.

Signed floating-point
number0*

Visibility depends on order options or device settings

Endress+Hauser

85Commissioning

Proline Teqwave MW 300 HART

10.5.5

Configuring the status input

The Status input submenu guides the user systematically through all the parameters thathave to be set for configuring the status input.
Navigation"Setup" menu → Status input 1 to n

‣ Status input 1 to n
Assign status input→  86

Terminal number→  86

Active level→  86

Terminal number→  86

Response time status input→  86

Terminal number→  86

Parameter overview with brief description
Parameter

Description

Selection / User interface /User entry

Factory setting

Assign status input

Select function for the status input.

• Off• Reset totalizer 1• Flow override

Off

Terminal number

Shows the terminal numbers used by thestatus input module.

• Not used
• 24-25 (I/O 2)• 22-23 (I/O 3)–

Active level

Define input signal level at which theassigned function is triggered.

• High• Low

HighResponse time status input

Define the minimum amount of time the
5 to 200 ms
input signal level must be present before theselected function is triggered.

86

50 ms

Endress+Hauser

Proline Teqwave MW 300 HART

Commissioning

10.5.6

Configuring the current output

The Current output wizard guides you systematically through all the parameters that haveto be set for configuring the current output.
Navigation"Setup" menu → Current output‣ Current output 1 to n
Terminal number→  87Signal mode→  87

Process variable current output→  87

Current range output→  88Lower range value output→  88Upper range value output→  88Fixed current→  88Damping current output→  88

Failure behavior current output→  88

Failure current→  88Parameter overview with brief description
Parameter

Prerequisite

Description

User interface /Selection / User
entry

Factory setting

Terminal number–

Shows the terminal numbers
used by the current output
module.

• Not used
• 26-27 (I/O 1)• 24-25 (I/O 2)• 22-23 (I/O 3)–

Signal mode–

Select the signal mode for thecurrent output.

• Active *• Passive *ActiveProcess variable current output

The Load rate option is only
available if the volume flow of
the medium is read in via theCurrent input 1 to n or thefieldbus.

Select the process variable forthe current output.

• Off• Total solids• Temperature• Electronics
temperature• Conductivity• Corrected
conductivity• Load rate *Total solids

Endress+Hauser

87Commissioning

Proline Teqwave MW 300 HART

Parameter

Prerequisite

Description

User interface /Selection / User
entry

Factory setting

Current range output–

Select current range forprocess value output andupper/lower level for alarm
signal.

Lower range value output

In Current span parameter(→  88), one of thefollowing options is selected:• 4...20 mA NE (3.8...20.5
mA)• 4...20 mA US (3.9...20.8
mA)• 4...20 mA (4...20.5 mA)• 0...20 mA (0...20.5 mA)

Enter lower range value for the Floating point
measured value range.
number with sign

0 %TS

Upper range value output

In Current span parameter(→  88), one of thefollowing options is selected:• 4...20 mA NE (3.8...20.5
mA)• 4...20 mA US (3.9...20.8
mA)• 4...20 mA (4...20.5 mA)• 0...20 mA (0...20.5 mA)

Enter upper range value forthe measured value range.

Floating point
number with sign

12 %TS

Fixed current

The Fixed current option is
selected in the Current spanparameter (→  88).

Defines the fixed output
current.

0 to 22.5 mA

22.5 mA

Damping current output

A process variable is selectedin the Assign current output
parameter (→  87) and oneof the following options isselected in the Current spanparameter (→  88):
• 4...20 mA NE (3.8...20.5
mA)• 4...20 mA US (3.9...20.8
mA)• 4...20 mA (4...20.5 mA)• 0...20 mA (0...20.5 mA)

Enter time constant for output
damping (PT1 element).
Damping reduces the effect offluctuations in the measured
value on the output signal.

0.0 to 999.9 s

1.0 s

Failure behavior current output

A process variable is selectedin the Assign current output
parameter (→  87) and oneof the following options isselected in the Current spanparameter (→  88):
• 4...20 mA NE (3.8...20.5
mA)• 4...20 mA US (3.9...20.8
mA)• 4...20 mA (4...20.5 mA)• 0...20 mA (0...20.5 mA)

Select output behavior in theevent of a device alarm.

• Min.
• Max.
• Last valid value• Actual value• Fixed value

Max.

Failure current

The Defined value option is
selected in the Failure modeparameter.

Enter current output value inalarm condition.

0 to 22.5 mA

22.5 mA*

88• 4...20 mA NE
Depends on country:(3.8...20.5 mA)• 4...20 mA NE• 4...20 mA US(3.8...20.5 mA)(3.9...20.8 mA)• 4...20 mA US• 4...20 mA (4...20.5(3.9...20.8 mA)
mA)• 0...20 mA (0...20.5
mA)• Fixed value

Visibility depends on order options or device settings

Endress+Hauser

Proline Teqwave MW 300 HART

Commissioning

10.5.7

Configuring the pulse/frequency/switch output

The Pulse/frequency/switch output wizard guides you systematically through all theparameters that can be set for configuring the selected output type.
Navigation"Setup" menu → Advanced setup → Pulse/frequency/switch output‣ Pulse/frequency/switch output
1 to n→  89Operating mode

Parameter overview with brief description
Parameter
Operating mode

Prerequisite

Description

If the Pulse option is selected,
the Load rate option must be
selected in the Assign pulse
output parameter.

Define the output as a pulse,
frequency or switch output.

Selection• Pulse• Frequency• Switch

Factory setting
Pulse

Configuring the pulse output
Navigation"Setup" menu → Pulse/frequency/switch output‣ Pulse/frequency/switch output
1 to nEndress+Hauser

Operating mode

→  90Terminal number→  90Signal mode

→  90Assign pulse output→  90

Pulse scaling→  90

Pulse width→  90

Failure mode

→  9089

Commissioning

Proline Teqwave MW 300 HART

Parameter overview with brief description
Parameter

Prerequisite

Description

Operating mode

If the Pulse option is selected,
the Load rate option must be
selected in the Assign pulse
output parameter.

Define the output as a pulse,
frequency or switch output.

• Pulse• Frequency• Switch

Pulse

Terminal number–

Shows the terminal numbers
used by the PFS output
module.

• Not used
• 24-25 (I/O 2)• 22-23 (I/O 3)–

Signal mode–

Select the signal mode for the
PFS output.

• Passive• Active *• Passive NE

Passive

Assign pulse output

The Load rate option is only
available if the volume flow of
the medium is read in via theCurrent input 1 to n or thefieldbus.
The Pulse option is selected in
Operating mode parameter.

Select process variable forpulse output.

• Off• Load rate *Off

Pulse scaling

The Pulse option is selected in
the Operating modeparameter (→  89) and aprocess variable is selected in
the Assign pulse output
parameter (→  90).

Enter quantity for measured
value at which a pulse is
output.

Positive floating
point number

Depends on country
and nominal
diameter

Pulse width

The Pulse option is selected in
the Operating modeparameter (→  89) and aprocess variable is selected in
the Assign pulse output
parameter (→  90).

Define time width of theoutput pulse.

0.05 to 2 000 ms

100 msFailure mode

The Pulse option is selected in
the Operating modeparameter (→  89) and aprocess variable is selected in
the Assign pulse output
parameter (→  90).

Select output behavior in theevent of a device alarm.

• Actual value• No pulses

No pulses*

Selection / Userinterface / User
entry

Factory setting

Visibility depends on order options or device settings

Configuring the frequency output
Navigation"Setup" menu → Pulse/frequency/switch output‣ Pulse/frequency/switch output
1 to n90

Operating mode

→  91

Terminal number→  91Signal mode

→  91

Assign frequency output→  91

Endress+Hauser

Proline Teqwave MW 300 HART

Commissioning

Minimum frequency value→  91

Maximum frequency value→  91

Measuring value at minimum
frequency→  92

Measuring value at maximum
frequency→  92

Failure mode

→  92

Failure frequency→  92Invert output signal→  92

Parameter overview with brief description
Parameter

Prerequisite

Description

Operating mode

If the Pulse option is selected,
the Load rate option must be
selected in the Assign pulse
output parameter.

Define the output as a pulse,
frequency or switch output.

• Pulse• Frequency• Switch

Pulse

Terminal number–

Shows the terminal numbers
used by the PFS output
module.

• Not used
• 24-25 (I/O 2)• 22-23 (I/O 3)–

Signal mode–

Select the signal mode for the
PFS output.

• Passive• Active *• Passive NE

Passive

Assign frequency output• The Frequency option is
Select process variable forselected in Operating mode frequency output.
parameter (→  89).
• The Load rate option is only
available if the volume flowof the medium is read in via
the Current input 1 to n or
the fieldbus.

• Off• Total solids• Temperature• Electronics
temperature• Conductivity• Corrected
conductivity• Load rate *Off

Minimum frequency value

The Frequency option is
selected in the Operating
mode parameter (→  89)
and a process variable isselected in the Assign
frequency output parameter(→  91).

Enter minimum frequency.

0.0 to 10 000.0 Hz

0.0 HzMaximum frequency value

The Frequency option is
selected in the Operating
mode parameter (→  89)
and a process variable isselected in the Assign
frequency output parameter(→  91).

Enter maximum frequency.

0.0 to 10 000.0 Hz

10 000.0 Hz

Endress+Hauser

Selection / Userinterface / User
entry

Factory setting

91Commissioning

Proline Teqwave MW 300 HART

Parameter

Prerequisite

Description

Selection / Userinterface / User
entry

Factory setting

Measuring value at minimum
frequency

The Frequency option is
selected in the Operating
mode parameter (→  89)
and a process variable isselected in the Assign
frequency output parameter(→  91).

Enter measured value forminimum frequency.

Signed floating-point
numberDepends on country
and nominal
diameter

Measuring value at maximum
frequency

The Frequency option is
selected in the Operating
mode parameter (→  89)
and a process variable isselected in the Assign
frequency output parameter(→  91).

Enter measured value for
maximum frequency.

Signed floating-point
numberDepends on country
and nominal
diameter

Failure mode

The Frequency option is
selected in the Operating
mode parameter (→  89)
and a process variable isselected in the Assign
frequency output parameter(→  91).

Select output behavior in theevent of a device alarm.

• Actual value• Defined value• 0 Hz

0 Hz

Failure frequency

In the Operating modeparameter (→  89), theFrequency option is selected,
in the Assign frequency
output parameter (→  91) a
process variable is selected,
and in the Failure modeparameter, the Defined value
option is selected.

Enter frequency output value
in alarm condition.

0.0 to 12 500.0 Hz

0.0 HzInvert output signal–

Invert the output signal.

• No
• Yes

No

*

92

Visibility depends on order options or device settings

Endress+Hauser

Proline Teqwave MW 300 HART

Commissioning

Configuring the switch output
Navigation"Setup" menu → Pulse/frequency/switch output‣ Pulse/frequency/switch output
1 to nOperating mode

→  93

Terminal number→  93Signal mode

→  93

Switch output function→  94

Assign diagnostic behavior→  94

Assign limit→  94

Assign status→  94

Switch-on value→  94

Switch-off value→  94

Switch-on delay→  94

Switch-off delay→  94

Failure mode

→  94

Parameter overview with brief description
Parameter

Prerequisite

Description

Operating mode

If the Pulse option is selected,
the Load rate option must be
selected in the Assign pulse
output parameter.

Define the output as a pulse,
frequency or switch output.

• Pulse• Frequency• Switch

Pulse

Terminal number–

Shows the terminal numbers
used by the PFS output
module.

• Not used
• 24-25 (I/O 2)• 22-23 (I/O 3)–

Signal mode–

Select the signal mode for the
PFS output.

• Passive• Active *• Passive NE

PassiveEndress+Hauser

Selection / Userinterface / User
entry

Factory setting

93Commissioning

Proline Teqwave MW 300 HART

Parameter

Prerequisite

Description

Selection / Userinterface / User
entry

Factory setting

Switch output function

The Switch option is selectedin the Operating modeparameter.

Select function for switch
output.

• Off• On• Diagnostic
behavior• Limit• Flow direction
check• Status

Off

Assign diagnostic behavior• In the Operating modeparameter, the Switch
option is selected.
• In the Switch output
function parameter, theDiagnostic behavior option
is selected.

The output is switched on(closed, conductive), if there isa pending diagnostic event of
the assigned behavioral
category.

• Alarm• Alarm or warning• Warning

Alarm

Assign limit• The Switch option is
selected in Operating modeparameter.
• The Limit option is selected
in Switch output function
parameter.
• The Load rate option is only
available if the volume flowof the medium is read in via
the Current input 1 to n or
the fieldbus.

Select the variable to monitor
in case the specified limit value
is exceeded. If a limit value is
exceeded, the output isswitched on (conductive).

• Off• Total solids• Temperature• Electronics
temperature• Conductivity• Corrected
conductivity• Load rate *• Totalizer 1 *Temperature

Assign status• The Switch option is
selected in Operating modeparameter.
• The Status option is
selected in Switch output
function parameter.

Select the device function forwhich to display the status. If
the switch on point is reached,
the output is switched on(closed, conductive).

• Off
Partially filled pipe
• Partially filled pipe detection
detection

Switch-on value• The Switch option is
selected in the Operating
mode parameter.
• The Limit option is selectedin the Switch output
function parameter.

Enter limit value for switch-on
point (process variable >switch-on value = closed,
conductive).

Floating point
number with sign

0 °C

Switch-off value• The Switch option is
selected in the Operating
mode parameter.
• The Limit option is selectedin the Switch output
function parameter.

Enter limit value for switch-off Floating point
point (process variable <number with sign
switch-off value = open,
nonconductive).

0 °C

Switch-on delay• The Switch option is
selected in the Operating
mode parameter.
• The Limit option is selectedin the Switch output
function parameter.

Enter a delay before the output 0.0 to 100.0 s
is switched on.

0.0 s

Switch-off delay• The Switch option is
selected in the Operating
mode parameter.
• The Limit option is selectedin the Switch output
function parameter.

Enter a delay before the output 0.0 to 100.0 s
is switched off.

0.0 s

Failure mode–

Select output behavior in theevent of a device alarm.

Open

*

94

• Actual status• Open• Closed

Visibility depends on order options or device settings

Endress+Hauser

Proline Teqwave MW 300 HART

Commissioning

10.5.8Configuring the relay output

The Relay output wizard guides the user systematically through all the parameters thathave to be set for configuring the relay output.
Navigation"Setup" menu → Relay output 1 to n

‣ Relay output 1 to n
Terminal number→  95

Relay output function→  95Assign flow direction check→  96

Assign limit→  96

Assign diagnostic behavior→  96

Assign status→  96

Switch-off value→  96

Switch-off delay→  96

Switch-on value→  96

Switch-on delay→  96

Failure mode

→  96

Switch state→  96

Powerless relay status→  96

Parameter overview with brief description
Parameter

Prerequisite

Description

User interface /Selection / User
entry

Factory setting

Terminal number–

Shows the terminal numbers
used by the relay output
module.

• Not used
• 24-25 (I/O 2)• 22-23 (I/O 3)–

Relay output function–

Select the function for therelay output.

• Closed• Open• Diagnostic
behavior• Limit• Flow direction
check• Status

ClosedEndress+Hauser

95

Commissioning

Proline Teqwave MW 300 HART

Parameter

Prerequisite

Description

Assign flow direction check

The Flow direction check
option is selected in the Relay
output function parameter.

Select process variable for flow
direction monitoring.

Assign limit• The Limit option is selected
in Relay output function
parameter.
• The Load rate option is only
available if the volume flowof the medium is read in via
the Current input 1 to n or
the fieldbus.

Select the variable to monitor
in case the specified limit value
is exceeded. If a limit value is
exceeded, the output isswitched on (conductive).

• Off• Total solids• Temperature• Electronics
temperature• Conductivity• Corrected
conductivity• Load rate *• Totalizer 1 *Temperature

Assign diagnostic behavior

In the Relay output function
parameter, the Diagnostic
behavior option is selected.

The output is switched on(closed, conductive), if there isa pending diagnostic event of
the assigned behavioral
category.

• Alarm• Alarm or warning• Warning

Alarm

Assign status

In the Relay output function
parameter, the Digital Output
option is selected.

Select the device function forwhich to display the status. If
the switch on point is reached,
the output is switched on(closed, conductive).

• Off
Off
• Partially filled pipe
detection

Switch-off value

The Limit option is selected in
the Relay output function
parameter.

Enter limit value for switch-off Floating point
point (process variable <number with sign
switch-off value = open,
nonconductive).

0 °C

Switch-off delay

In the Relay output function
parameter, the Limit option is
selected.

Enter a delay before the output 0.0 to 100.0 s
is switched off.

0.0 s

Switch-on value

The Limit option is selected in
the Relay output function
parameter.

Enter measured value for theswitch-on point.

0 °C

Switch-on delay

In the Relay output function
parameter, the Limit option is
selected.

Enter a delay before the output 0.0 to 100.0 s
is switched on.

0.0 s

Failure mode–

Select output behavior in theevent of a device alarm.

• Actual status• Open• ClosedOpen

Switch state–

Indicates the current switch
state of the output.

• Open• Closed–

Powerless relay status–

Select quiescent state for relay. • Open• Closed*User interface /Selection / User
entry

Factory setting

Off

Floating point
number with sign

Open

Visibility depends on order options or device settings

10.5.9Configuring the local display

The Display wizard guides you systematically through all the parameters that canconfigured for configuring the local display.
Navigation"Setup" menu → Display‣ Display96

Endress+Hauser

Proline Teqwave MW 300 HART

Commissioning

Format display→  97

Value 1 display→  970% bargraph value 1→  97

100% bargraph value 1→  97

Value 2 display→  98

Value 3 display→  980% bargraph value 3→  98

100% bargraph value 3→  98

Value 4 display→  98Parameter overview with brief description
Parameter

Prerequisite

Description

Factory setting• 1 value, max. size
• 1 bargraph + 1value• 2 values• 1 value large + 2
values• 4 values

1 value, max. size

Format display

A local display is provided.

Value 1 display• A local display is provided.
Select the measured value that
• The Load rate option is only is shown on the local display.
available if the volume flowof the medium is read in via
the Current input 1 to n or
the fieldbus.

• Total solids• Temperature• Electronics
temperature• Conductivity• Corrected
conductivity• Load rate *• Totalizer 1 *• Current output 1 *• Current output 2 *• Current output 3 *• Current output 4 *Total solids

0% bargraph value 1

A local display is provided.

Enter 0 % value for bar graphdisplay.

Signed floating-point
number

0 %TS100% bargraph value 1

A local display is provided.

Enter 100 % value for bar
graph display.

Signed floating-point
numberDepends on country
and nominal
diameterEndress+Hauser

Select how measured values
are shown on the display.

Selection / User
entry

97

Commissioning

Parameter

Proline Teqwave MW 300 HART

Prerequisite

Description

Selection / User
entry

Factory setting

Value 2 display• A local display is provided.
Select the measured value that
• The Load rate option is only is shown on the local display.
available if the volume flowof the medium is read in via
the Current input 1 to n or
the fieldbus.

• None• Total solids• Temperature• Electronics
temperature• Conductivity• Corrected
conductivity• Load rate *• Totalizer 1 *• Current output 1 *• Current output 2 *• Current output 3 *• Current output 4 *Value 3 display• A local display is provided.
Select the measured value that
• The Load rate option is only is shown on the local display.
available if the volume flowof the medium is read in via
the Current input 1 to n or
the fieldbus.

For the picklist, see
None
Value 1 display
parameter (→  97)

0% bargraph value 3

A selection was made in the
Value 3 display parameter.

Enter 0 % value for bar graphdisplay.

Signed floating-point
number0100% bargraph value 3

A selection was made in the
Value 3 display parameter.

Enter 100 % value for bar
graph display.

Signed floating-point
number0

Value 4 display• A local display is provided.
Select the measured value that
• The Load rate option is only is shown on the local display.
available if the volume flowof the medium is read in via
the Current input 1 to n or
the fieldbus.

*

98

None

For the picklist, see
None
Value 1 display
parameter (→  97)

Visibility depends on order options or device settings

Endress+Hauser

Proline Teqwave MW 300 HART

Commissioning

10.5.10 "Total solids commissioning" wizard
The Total solids commissioning wizard is used to perform the basic settings for adjusting
the measured value based on a reference value.
Wizard description →  130.
Navigation"Setup" menu → Total solids commissioning‣ Total solids commissioning

10.5.11 "Total solids adjustment" submenu
Using the Total solids adjustment submenu, you can call up the wizards for adjusting themeasured value based on a reference value.
Description and access to the wizards →  130
Navigation"Setup" menu → Total solids adjustment‣ Total solids adjustment

Endress+Hauser

99Commissioning

Proline Teqwave MW 300 HART

10.6

Advanced settings

The Advanced setup submenu with its submenus contains parameters for specificsettings.
Navigation to the "Advanced setup" submenu
XXXXXXXXX

20.50
mA
1.

Main menu
Display language
EnglishOperation
Setup
Main menu
Operation
Setup

2.

Diagnostic/ ../Setup
Device tag

3.

XXXXXXXXXXXXXXXXXX/ ../Setup
XXXXXXXXXXXXXXXXXX
Advanced setup

4.

/ ../Advanced setup
Ent. access code

5.

Totalizer
Display****
A0053791-EN

The number of submenus and parameters can vary depending on the device version
and the available application packages. These submenus and their parameters are
explained in the Special Documentation for the device and not in Operating
Instructions.
For detailed information on the parameter descriptions of the application packages,
see the Special Documentation for the device. →  192
Navigation"Setup" menu → Advanced setup‣ Advanced setup

100

Enter access code→  101‣ Totalizer 1→  102‣ Display→  103

Endress+Hauser

Proline Teqwave MW 300 HART

10.6.1

Commissioning‣ WLAN settings→  107‣ Heartbeat setup→  108‣ Configuration backup→  109‣ Administration→  111

Using the parameter to enter the access code

Navigation"Setup" menu → Advanced setup
Parameter overview with brief description
Parameter
Enter access code

Endress+Hauser

Description
Enter access code to disable write protection of parameters.

User entry
Max. 16-digit character string comprisingnumbers, letters and special characters

101Commissioning

Proline Teqwave MW 300 HART

10.6.2

Configuring the totalizer

The totalizer is used to calculate the total load rate. The Load rate option is only
available if the volume flow of the medium is read in via the Current input 1 to n.
In the "Totalizer 1 to n" submenu, you can configure the totalizer.
Navigation"Setup" menu → Advanced setup → Totalizer 1 to n

‣ Totalizer 1
Assign process variable 1→  102

Process variable unit 1→  102

Totalizer 1 operation mode→  102

Totalizer 1 failure behavior→  102

Parameter overview with brief description
Parameter

Prerequisite

Assign process variable 1

The Load rate option is only
available if the volume flow of
the medium is read in via theCurrent input 1 to n or thefieldbus.

Process variable unit 1

Description

Factory setting• Off• Load rate *Off

A process variable is selected
Select the unit for the process
in the Assign process variable variable of the totalizer.
parameter (→  102) of the
Totalizer 1 to n submenu.

Unit choose list

Depends on country

Totalizer 1 operation mode

A process variable is selectedin the Assign process variable
parameter (→  102) of the
Totalizer 1 to n submenu.

• Net• Forward• Reverse

Net

Totalizer 1 failure behavior

A process variable is selected
Select totalizer behavior in thein the Assign process variable event of a device alarm.
parameter (→  102) of the
Totalizer 1 to n submenu.

• Hold• Continue• Last valid value +continue

Hold

*

102

Select process variable fortotalizer.

Selection

Select totalizer operationmode, e.g. only totalize
forward flow or only totalize
reverse flow.

Visibility depends on order options or device settings

Endress+Hauser

Proline Teqwave MW 300 HART

Commissioning

10.6.3Carrying out additional display configurations

In the Display submenu you can set all the parameters associated with the configurationof the local display.
Navigation"Setup" menu → Advanced setup → Display‣ Display

Endress+Hauser

Format display→  104

Value 1 display→  104

0% bargraph value 1→  104

100% bargraph value 1→  104

Decimal places 1→  104

Value 2 display→  104

Decimal places 2→  104

Value 3 display→  104

0% bargraph value 3→  104

100% bargraph value 3→  104

Decimal places 3→  105

Value 4 display→  105

Decimal places 4→  105

Display language→  105

Display interval→  105

Display damping→  105

Header→  105

Header text

→  105

Separator→  106

Backlight→  106103

Commissioning

Proline Teqwave MW 300 HART

Parameter overview with brief description
Parameter

Prerequisite

Description

Factory setting• 1 value, max. size
• 1 bargraph + 1value• 2 values• 1 value large + 2
values• 4 values

1 value, max. size

Format display

A local display is provided.

Value 1 display• A local display is provided.
Select the measured value that
• The Load rate option is only is shown on the local display.
available if the volume flowof the medium is read in via
the Current input 1 to n or
the fieldbus.

• Total solids• Temperature• Electronics
temperature• Conductivity• Corrected
conductivity• Load rate *• Totalizer 1 *• Current output 1 *• Current output 2 *• Current output 3 *• Current output 4 *Total solids

0% bargraph value 1

A local display is provided.

Enter 0 % value for bar graphdisplay.

Signed floating-point
number

0 %TS100% bargraph value 1

A local display is provided.

Enter 100 % value for bar
graph display.

Signed floating-point
numberDepends on country
and nominal
diameter

Decimal places 1

A measured value is specified
in the Value 1 display
parameter.

Select the number of decimal
places for the display value.

• x
• x.x
• x.xx• x.xxx• x.xxxx

x.xx

Value 2 display• A local display is provided.
Select the measured value that
• The Load rate option is only is shown on the local display.
available if the volume flowof the medium is read in via
the Current input 1 to n or
the fieldbus.

• None• Total solids• Temperature• Electronics
temperature• Conductivity• Corrected
conductivity• Load rate *• Totalizer 1 *• Current output 1 *• Current output 2 *• Current output 3 *• Current output 4 *None

Decimal places 2

A measured value is specified
in the Value 2 display
parameter.

• x
• x.x
• x.xx• x.xxx• x.xxxx

x.xx

Value 3 display• A local display is provided.
Select the measured value that
• The Load rate option is only is shown on the local display.
available if the volume flowof the medium is read in via
the Current input 1 to n or
the fieldbus.

For the picklist, see
None
Value 1 display
parameter (→  97)

0% bargraph value 3

A selection was made in the
Value 3 display parameter.

Enter 0 % value for bar graphdisplay.

Signed floating-point
number0100% bargraph value 3

A selection was made in the
Value 3 display parameter.

Enter 100 % value for bar
graph display.

Signed floating-point
number0

104

Select how measured values
are shown on the display.

Selection / User
entry

Select the number of decimal
places for the display value.

Endress+Hauser

Proline Teqwave MW 300 HART

Parameter

Commissioning

Prerequisite

Description

Decimal places 3

A measured value is specified
in the Value 3 display
parameter.

Select the number of decimal
places for the display value.

Value 4 display• A local display is provided.
Select the measured value that
• The Load rate option is only is shown on the local display.
available if the volume flowof the medium is read in via
the Current input 1 to n or
the fieldbus.

For the picklist, see
None
Value 1 display
parameter (→  97)

Decimal places 4

A measured value is specified
in the Value 4 display
parameter.

Select the number of decimal
places for the display value.

• x
• x.x
• x.xx• x.xxx• x.xxxx

x.xx

Display language

A local display is provided.

Set display language.

• English• Deutsch• Français• Español• Italiano• Nederlands• Portuguesa• Polski• русский язык(Russian)• Svenska• Türkçe• 中文 (Chinese)• 日本語 (Japanese)• 한국어 (Korean)• čeština (Czech)

English(alternatively, theordered language ispreset in the device)

Display interval

A local display is provided.

Set time measured values areshown on display if display
alternates between values.

1 to 10 s

5s

Display damping

A local display is provided.

Set display reaction time tofluctuations in the measured
value.

0.0 to 999.9 s

0.0 s

Header

A local display is provided.

Select header contents on local • Device tag
display.
• Free text

Device tag

Header text

The Free text option is
selected in the Header
parameter.

Enter display header text.

------------

Endress+Hauser

Selection / User
entry
• x
• x.x
• x.xx• x.xxx• x.xxxx

Max. 12 characters,such as letters,
numbers or special
characters (e.g. @,%, /)Factory setting
x.xx

105

Commissioning

Parameter

Proline Teqwave MW 300 HART

Prerequisite

Description

Separator

A local display is provided.

Backlight

One of the following
Switch the local display
conditions is met:
backlight on and off.
• Order code for "Display;
operation", option F "4-line,
illum.; touch control"• Order code for "Display;
operation", option G "4-line,
illum.; touch control+WLAN"• Order code for "Display;
operation", option O "Remote
display 4-line illuminated;
10m/30ft cable; touch
control"*

106

Select decimal separator for
displaying numerical values.

Selection / User
entry

Factory setting• . (point)• , (comma). (point)• Disable• Enable

EnableVisibility depends on order options or device settings

Endress+Hauser

Proline Teqwave MW 300 HART

Commissioning

10.6.4WLAN configuration

The WLAN Settings submenu guides the user systematically through all the parametersthat have to be set for the WLAN configuration.
Navigation"Setup" menu → Advanced setup → WLAN settings‣ WLAN settings
WLAN→  107

WLAN mode→  107

SSID name→  107

Network security→  108

Security identification→  108

User name→  108

WLAN password→  108

WLAN IP address→  108

WLAN MAC address→  108

WLAN passphrase→  108

Assign SSID name→  108

SSID name→  108

Connection state→  108

Received signal strength→  108

Parameter overview with brief description
Parameter

Prerequisite

Description

Selection / User
entry / Userinterface

Factory setting

WLAN

–

Switch WLAN on and off.

• Disable• Enable

EnableWLAN mode–

Select WLAN mode.

• WLAN accesspoint• WLAN Client

WLAN access point

SSID name

The client is activated.

Enter the user-defined SSID
name (max. 32 characters).

––

Endress+Hauser

107Commissioning

Proline Teqwave MW 300 HART

Parameter

Prerequisite

Description

Selection / User
entry / Userinterface

Factory setting

Network security–

Select the security type of the
WLAN network.

• Unsecured• WPA2-PSK• EAP-PEAP withMSCHAPv2 *• EAP-PEAP
MSCHAPv2 noserver authentic. *• EAP-TLS *WPA2-PSK

Security identification–

Select security settings anddownload these settings via
menu Data management >Security > WLAN.

• Trusted issuer
certificate• Device certificate• Device private key–

User name–Enter user name.

––

WLAN password–

Enter WLAN password.

––

WLAN IP address–Enter IP address of the WLAN
interface of the device.

4 octet: 0 to 255 (inthe particular octet)

192.168.1.212

WLAN MAC address–Enter MAC address of the
WLAN interface of the device.

Unique 12-digit
character string
comprising lettersand numbers

Each measuring
device is given an
individual address.

WLAN passphrase

The WPA2-PSK option is
selected in the Security typeparameter.

Enter the network key (8 to 32 8 to 32-digit
characters).
character string
comprising numbers,
The network key
supplied with the device letters and special
characters (without
should be changed
spaces)during commissioningSerial number of themeasuring device(e.g. L100A802000)

for security reasons.
Assign SSID name–

Select which name will be used • Device tag
for SSID: device tag or user• User-defined
defined name.

SSID name• The User-defined option is
selected in the Assign SSID
name parameter.
• The WLAN access point
option is selected in the
WLAN mode parameter.

Enter the user-defined SSID
name (max. 32 characters).

Connection state–

Displays the connection status. • Connected• Not connected

Not connected

Received signal strength–

Shows the received signal
strength.

High

*



User-defined

Max. 32-digit
character string
comprising numbers,
The user-defined SSID
letters and specialname may only be
assigned once. If the SSID charactersname is assigned morethan once, the devices
can interfere with one
another.

• Low• Medium• HighVisibility depends on order options or device settings

10.6.5

Heartbeat Technology application package

For detailed information on the parameter descriptions of the application packages,
see the Special Documentation for the device. →  192
Navigation"Setup" menu → Advanced setup → Heartbeat setup

108

Endress+Hauser

Proline Teqwave MW 300 HART

Commissioning

10.6.6

Configuration management

After commissioning, you can save the current device configurationor restore the previous
device configuration. The device configuration is managed via the Configuration
management parameter.
Navigation"Setup" menu → Advanced setup → Configuration backup‣ Configuration backup
Operating time→  109

Last backup→  109

Configuration management→  109

Backup state→  109

Comparison result→  109

Parameter overview with brief description
Parameter

Description

User interface / Selection

Factory setting

Operating time

Indicates how long the device has been in
operation.

Days (d), hours (h), minutes(m) and seconds (s)–Last backup

Shows when the last data backup was saved
to HistoROM backup.

Days (d), hours (h), minutes(m) and seconds (s)–Configuration management

Select action for managing the device data in • Cancel
the HistoROM backup.
• Execute backup• Restore *• Compare *• Clear backup data

CancelBackup stateShows the current status of data saving or
restoring.

• None• Backup in progress• Restoring in progress• Delete in progress• Compare in progress• Restoring failed• Backup failed

None

Comparison result

Comparison of current device data with
HistoROM backup.

• Settings identical• Settings not identical• No backup available• Backup settings corrupt• Check not done
• Dataset incompatible

Check not done*

Visibility depends on order options or device settings

Endress+Hauser

109

Commissioning

Proline Teqwave MW 300 HART

Function scope of the "Configuration management" parameter
Options

Description

CancelNo action is executed and the user exits the parameter.

Execute backup

A backup copy of the current device configuration is saved from the HistoROM
backup to the memory of the device. The backup copy includes the transmitter dataof the device.

RestoreThe last backup copy of the device configuration is restored from the device
memory to the device's HistoROM backup. The backup copy includes thetransmitter data of the device.

CompareThe device configuration saved in the device memory is compared with the current
device configuration of the HistoROM backup.

Clear backup data

The backup copy of the device configuration is deleted from the memory of thedevice.

HistoROM backup
A HistoROM is a "non-volatile" device memory in the form of an EEPROM.
While this action is in progress, the configuration cannot be edited via the local
display and a message on the processing status appears on the display.

110

Endress+Hauser

Proline Teqwave MW 300 HART

Commissioning

10.6.7

Using parameters for device administration

The Administration submenu systematically guides the user through all the parametersthat can be used for device administration purposes.
Navigation"Setup" menu → Advanced setup → Administration‣ Administration‣ Define access code→  111‣ Reset access code→  111

Device reset→  112Using the parameter to define the access code
Complete this wizard to specify an access code for the Maintenance role.
Navigation"Setup" menu → Advanced setup → Administration → Define access code

‣ Define access code
Define access code→  111

Confirm access code→  111

Parameter overview with brief description
Parameter

Description

User entry

Define access code

Specify an access code that is required to obtain the accessrights for the Maintenance role.

Max. 16-digit character string comprisingnumbers, letters and special characters

Confirm access code

Confirm the access code entered for the Maintenance role.

Max. 16-digit character string comprisingnumbers, letters and special characters

Using the parameter to reset the access code
Navigation"Setup" menu → Advanced setup → Administration → Reset access code

‣ Reset access code

Endress+Hauser

Operating time→  112

Reset access code→  112111

Commissioning

Proline Teqwave MW 300 HART

Parameter overview with brief description
Parameter

Description

User interface / User entry

Factory setting

Operating time

Indicates how long the device has been in
operation.

Days (d), hours (h), minutes(m) and seconds (s)–Reset access code

Enter the code provided by Endress+Hauser
Technical Support to reset the Maintenancecode.

Character string comprisingnumbers, letters and special
characters

0x00

a reset code, contact your
 For
Endress+Hauser service organization.
The reset code can only be entered via:
• Web browser• DeviceCare, FieldCare (via CDI-RJ45
service interface)• Fieldbus

Using the parameter to reset the device
Navigation"Setup" menu → Advanced setup → Administration
Parameter overview with brief description
Parameter
Device reset*

112

Description
Reset the device configuration - either
entirely or in part - to a defined state.

Selection• Cancel• To delivery settings• Restart device• Restore S-DAT backup *Factory setting
CancelVisibility depends on order options or device settings

Endress+Hauser

Proline Teqwave MW 300 HART

Commissioning

10.7

Simulation

Via the Simulation submenu, it is possible to simulate various process variables in theprocess and the device alarm mode and verify downstream signal chains (switching valvesor closed-control loops). The simulation can be performed without a real measurement (noflow of medium through the device).
Navigation"Diagnostics" menu → Simulation‣ Simulation

Endress+Hauser

Assign simulation process variable→  114

Process variable value→  114

Current input 1 to n simulation→  114

Value current input 1 to n

→  114

Status input 1 to n simulation→  114

Input signal level 1 to n

→  114

Current output 1 to n simulation→  114

Current output value→  114Frequency output 1 to n simulation→  114Frequency output 1 to n value→  114

Pulse output simulation 1 to n

→  114

Pulse value 1 to n

→  114

Switch output simulation 1 to n

→  114

Switch state 1 to n

→  114

Relay output 1 to n simulation→  114

Switch state 1 to n

→  114

Device alarm simulation→  114

Diagnostic event category→  115

Diagnostic event simulation→  115113

Commissioning

Proline Teqwave MW 300 HART

Parameter overview with brief description
Parameter

Prerequisite

Description

Assign simulation process variable

The Load rate option is only
available if the volume flow of
the medium is read in via theCurrent input 1 to n or thefieldbus.

Select a process variable forthe simulation process that is
activated.

• Off• Load rate *• Total solids• Temperature• Electronics
temperature• Conductivity• Corrected
conductivity

Off

Process variable value–Enter the simulation value forthe selected process variable.

Signed floating-point
number0

Current input 1 to n simulation–

Switch simulation of thecurrent input on and off.

• Off
• On

Off

Value current input 1 to n

In the Current input 1 to nsimulation parameter, the On
option is selected.

Enter the current value forsimulation.

0 to 22.5 mA

0 mA

Status input 1 to n simulation–

Switch simulation of the status • Off
input on and off.
• On

Input signal level 1 to n

In the Status input simulation Select the signal level for theparameter, the On option issimulation of the status input.
selected.

• High• Low

HighCurrent output 1 to n simulation–

Switch the simulation of thecurrent output on and off.

• Off
• On

Off

Current output value

In the Current output 1 to nsimulation parameter, the On
option is selected.

Enter the current value forsimulation.

3.59 to 22.5 mA

3.59 mA

Frequency output 1 to n simulation

In the Operating modeparameter, the Frequency
option is selected.

Switch the simulation of thefrequency output on and off.

• Off
• On

Off

Frequency output 1 to n value

In the Frequency simulation
1 to n parameter, the On
option is selected.

Enter the frequency value forthe simulation.

0.0 to 12 500.0 Hz

0.0 HzPulse output simulation 1 to n

In the Operating modeparameter, the Pulse option is
selected.

Set and switch off the pulse
output simulation.

• Off• Fixed value• Down-counting
value

Off

For Fixed value option:
 Pulsewidth parameter

Selection / User
entry

Factory settingOff(→  90) defines thepulse width of the pulses
output.
Pulse value 1 to n

In the Pulse output
simulation 1 to n parameter,
the Down-counting value
option is selected.

Enter the number of pulses for
simulation.

0 to 65 535

0

Switch output simulation 1 to n

In the Operating modeparameter, the Switch option
is selected.

Switch the simulation of theswitch output on and off.

• Off
• On

Off

Switch state 1 to n–

Select the status of the status
output for the simulation.

• Open• Closed

Open

Relay output 1 to n simulation–

Switch simulation of the relay
output on and off.

• Off
• On

Off

Switch state 1 to n

The On option is selected in
the Switch output simulation
1 to n parameter parameter.

Select status of the relay
output for the simulation.

• Open• ClosedOpen

Device alarm simulation–

Switch the device alarm on and • Off
off.
• On

114

Off

Endress+Hauser

Proline Teqwave MW 300 HART

Parameter

Commissioning

Prerequisite

Description

Selection / User
entry

Factory setting

Diagnostic event category–

Select a diagnostic event
category.

• Sensor• Electronics• Configuration• ProcessProcess

Diagnostic event simulation–

Select a diagnostic event tosimulate this event.

• Off• Diagnostic event
picklist (depends
on the category
selected)

Off*

Visibility depends on order options or device settings

Endress+Hauser

115

Commissioning

Proline Teqwave MW 300 HART

10.8

Protecting settings from unauthorized access

The following write protection options exist in order to protect the configuration of themeasuring device from unintentional modification:• Protect access to parameters via access code →  116• Protect access to local operation via key locking →  59
• Protect access to measuring device via write protection switch →  117

10.8.1

Write protection via access code

The effects of the user-specific access code are as follows:• Via local operation, the parameters for the measuring device configuration are writeprotected and their values can no longer be changed.
• Device access is protected via the Web browser, as are the parameters for the measuring
device configuration.
• Device access is protected via FieldCare or DeviceCare (via CDI-RJ45 service interface), as
are the parameters for the measuring device configuration.
Defining the access code via the local display
1.

Navigate to the Define access code parameter (→  111).

2.

Maximum of 16-digit character string comprising numbers, letters and special
characters as the access code.

3.

Enter the access code again in the Confirm access code parameter (→  111) toconfirm.
 The  symbol appears in front of all write-protected parameters.
• Disabling parameter write protection via access code →  59.
• If the access code is lost: Resetting the access code →  117.
• The user role with which the user is currently logged in is displayed in Accessstatus parameter.
• Navigation path: Operation → Access status• User roles and their access rights →  58

• The device automatically locks the write-protected parameters again if a key is notpressed for 10 minutes in the navigation and editing view.
• The device locks the write-protected parameters automatically after 60 s if the user skips
back to the operational display mode from the navigation and editing view.
Parameters which can always be modified via the local display
Certain parameters that do not affect the measurement are excepted from parameter write
protection via the local display. Despite the user-specific access code, they can always be
modified, even if the other parameters are locked.
Parameters for configuring
the language

Parameters for configuring
the local display

Parameters for configuring
the totalizer

↓

↓

↓

Display language

Format display

Control Totalizer

Contrast display

Preset value

Display interval

Reset all totalizers

Defining the access code via the web browser
1.

116

Navigate to the Define access code parameter (→  111).

Endress+Hauser

Proline Teqwave MW 300 HART

Commissioning

2.

Define a 16-digit (max.) numeric code as the access code.

3.

Enter the access code again in the Confirm access code parameter (→  111) toconfirm.
 The web browser switches to the login page.
• Disabling parameter write protection via access code →  59.
• If the access code is lost: Resetting the access code →  117.
• The Access status parameter shows which user role the user is currently logged inwith.
• Navigation path: Operation → Access status• User roles and their access rights →  58

If no action is performed for 10 minutes, the web browser automatically returns to thelogin page.
Resetting the access code
If you misplace the user-specific access code, it is possible to reset the code to the factory
setting. A reset code must be entered for this purpose. The user-specific access code canthen be defined again afterwards.
Via Web browser, FieldCare, DeviceCare (via CDI-RJ45 service interface), fieldbus
You can only obtain a reset code from your local Endress+Hauser service organization.
The code must be calculated explicitly for every device.
1.

Note down the serial number of the device.

2.

Read off the Operating time parameter.

3.

Contact the local Endress+Hauser service organization and tell them the serial
number and the operating time.
 Get the calculated reset code.

4.

Enter the reset code in the Reset access code parameter (→  112).
 The access code has been reset to the factory setting 0000. It can be redefined→  116.
For IT security reasons, the calculated reset code is only valid for 96 hours from thespecified operating time and for the specific serial number. If you cannot return to thedevice within 96 hours, you should either increase the operating time you read out by
a few days or switch off the device.

10.8.2

Write protection via write protection switch

Unlike parameter write protection via a user-specific access code, this allows the user tolock write access to the entire operating menu - apart from the "Contrast display"parameter.
The parameter values are now read only and cannot be edited any more (exception"Contrast display" parameter):• Via local display• Via HART protocol

Endress+Hauser

117Commissioning

Proline Teqwave MW 300 HART

1.
OFF ON
1
2
3
4

A0029630

Setting the write protection (WP) switch on the main electronics module to the ON
position enables hardware write protection.
 In the Locking status parameter, the Hardware locked option is displayed→  119. In addition, on the local display the  symbol appears in front of theparameters in the header of the operational display and in the navigation view.

XXXXXXXXX

20.50
XX

XX
A0029425

2.

118

Setting the write protection (WP) switch on the main electronics module to the OFF
position (factory setting) disables hardware write protection.
 No option is displayed in the Locking status parameter →  119. On the local
display, the  symbol disappears from in front of the parameters in the headerof the operational display and in the navigation view.

Endress+Hauser

Proline Teqwave MW 300 HART

Operation

11

Operation

11.1

Reading off the device locking status

Device active write protection: Locking status parameter
Operation → Locking status
Function scope of the "Locking status" parameter
Options

Description

None

The access authorization displayed in the Access status parameter applies→  58. Only appears on local display.

Hardware locked

The DIP switch for hardware locking is activated on the PCB board. This locks write
access to the parameters (e.g. via local display or operating tool) →  117.

Temporarily locked

Write access to the parameters is temporarily locked on account of internal
processes running in the device (e.g. data upload/download, reset, etc.). Once theinternal processing has been completed, the parameters can be changed once
again.

11.2

Adjusting the operating language

Detailed information:• To configure the operating language• For information on the operating languages supported by the measuring device→  181

11.3

Configuring the display

Detailed information:• On the basic settings for the local display →  96
• On the advanced settings for the local display →  103

11.4

Adapting the measuring device to the process
conditions

The following are available for this purpose:• Basic settings using theSetup menu (→  81)• Advanced settings using theAdvanced setup submenu (→  100)

Endress+Hauser

119Operation

Proline Teqwave MW 300 HART

11.5

Reading off measured values

With the Measured values submenu, it is possible to read all the measured values.
Navigation"Diagnostics" menu → Measured values‣ Measured values

11.5.1‣ Process variables→  120‣ Input values→  121‣ Output values→  122‣ Totalizer→  124"Process variables" submenu

The contains all the parameters needed to display the current measured values for eachprocess variable.
Navigation"Diagnostics" menu → Measured values → Process variables‣ Process variables
Total solids→  120

Temperature→  120

Electronics temperature→  120

Conductivity→  121

Corrected conductivity→  121

Load rate→  121

Parameter overview with brief description
Parameter

Prerequisite

Description

User interface

Total solids–

Shows total solids (fraction of total
weight or concentration per volumeunit).

Signed floating-point
numberTemperature–

Shows the medium temperature
currently measured.

Signed floating-point
numberElectronics temperature–

Shows the electronics temperature
currently measured.

Signed floating-point
number120

Endress+Hauser

Proline Teqwave MW 300 HART

Operation

Parameter

Prerequisite

Description

User interface

Conductivity–

Shows the conductivity currently
measured.

Floating-point number

Corrected conductivity–

Shows the conductivity measured
compensated for temperature.

Floating-point number

Load rate

The volume flow of the medium is readin via the Current input 1 to n or thefieldbus.

Shows the total solids flow rate.

Signed floating-point
number

11.5.2"Input values" submenu

The Input values submenu guides you systematically to the individual input values.
Navigation"Diagnostics" menu → Measured values → Input values‣ Input values‣ Current input 1 to n

→  121‣ Status input 1 to n

→  121

Input values of current input
The Current input 1 to n submenu contains all the parameters needed to display thecurrent measured values for every current input.
Navigation"Diagnostics" menu → Measured values → Input values → Current input 1 to n

‣ Current input 1 to n
Measured values 1 to n

→  121

Measured current 1 to n

→  121

Parameter overview with brief description
Parameter

Description

User interface

Measured values 1 to n

Displays the current input value.

Signed floating-point number

Measured current 1 to n

Displays the current value of the current input.

0 to 22.5 mA

Input values of status input
The Status input 1 to n submenu contains all the parameters needed to display thecurrent measured values for every status input.

Endress+Hauser

121

Operation

Proline Teqwave MW 300 HART

Navigation"Diagnostics" menu → Measured values → Input values → Status input 1 to n

‣ Status input 1 to n
→  122

Value status input

Parameter overview with brief description
Parameter
Value status input

Description
Shows the current input signal level.

11.5.3User interface• High• Low

Output values

The Output values submenu contains all the parameters needed to display the current
measured values for every output.
Navigation"Diagnostics" menu → Measured values → Output values‣ Output values‣ Current output 1 to n

→  122‣ Pulse/frequency/switch output→  123‣ Relay output 1 to n

→  123

1 to nOutput values of current output
The Value current output submenu contains all the parameters needed to display thecurrent measured values for every current output.
Navigation"Diagnostics" menu → Measured values → Output values → Value current output 1 to n

‣ Current output 1 to n

122

Output current→  123

Measured current→  123

Endress+Hauser

Proline Teqwave MW 300 HART

Operation

Parameter overview with brief description
Parameter

Description

User interface

Output current

Displays the current value currently calculated for the current
output.

3.59 to 22.5 mA

Measured current

Displays the current value currently measured for the current
output.

0 to 30 mA

Output values for pulse/frequency/switch output
The Pulse/frequency/switch output 1 to n submenu contains all the parameters needed
to display the current measured values for every pulse/frequency/switch output.
Navigation"Diagnostics" menu → Measured values → Output values → Pulse/frequency/switch output
1 to n‣ Pulse/frequency/switch output
1 to nOutput frequency→  123

Pulse output→  123

Switch state→  123

Parameter overview with brief description
Parameter

Prerequisite

Description

Output frequency

In the Operating mode parameter, theFrequency option is selected.

Displays the value currently measured
for the frequency output.

Pulse output

The Pulse option is selected in the
Displays the pulse frequency currently
Operating mode parameter parameter. output.

Positive floating-point
number

Switch state

The Switch option is selected in the
Operating mode parameter.

• Open• Closed

Displays the current switch output
status.

User interface
0.0 to 12 500.0 Hz

Output values for relay output
The Relay output 1 to n submenu contains all the parameters needed to display thecurrent measured values for every relay output.
Navigation"Diagnostics" menu → Measured values → Output values → Relay output 1 to n

‣ Relay output 1 to n

Endress+Hauser

Switch state→  124

Switch cycles→  124

Max. switch cycles number→  124

123

Operation

Proline Teqwave MW 300 HART

Parameter overview with brief description
Parameter

Description

User interface

Switch state

Indicates the current switch state of the output.

• Open• Closed

Switch cycles

Shows number of all performed switch cycles.

Positive integer

Max. switch cycles number

Shows the maximal number of guaranteed switch cycles.

Positive integer

11.5.4"Totalizer" submenu

The Totalizer submenu contains all the parameters needed to display the current
measured values for every totalizer.
Navigation"Diagnostics" menu → Measured values → Totalizer‣ Totalizer
Totalizer 1 value→  124

Totalizer 1 overflow→  124

Parameter overview with brief description
Parameter

Description

User interface

Totalizer 1 value

Displays the current totalizer counter value.

Signed floating-point number

Totalizer 1 overflow

Displays the current totalizer overflow.

Integer with sign

124

Endress+Hauser

Proline Teqwave MW 300 HART

Operation

11.6

Performing a totalizer reset

The totalizers are reset in the Operation submenu:• Control Totalizer• Reset all totalizers
Navigation"Operation" menu → Totalizer handling‣ Totalizer handlingControl Totalizer 1 to n

→  125

Preset value 1 to n

→  125

Totalizer value 1 to n

→  125

Reset all totalizers→  125

Parameter overview with brief description
Parameter

Prerequisite

Description

Selection / User
entry / Userinterface

Factory setting

Totalizer 1 control

A process variable is selected
Operate the totalizer.
in the Assign process variable
parameter (→  102) of the
Totalizer 1 to n submenu.

• Totalize• Reset + hold
• Preset + hold
• Reset + totalize• Preset + totalize• HoldPreset value 1

A process variable is selected
Specify start value for totalizer. Signed floating-pointin the Assign process variable Dependency
numberparameter (→  102) of the
The unit of the selected
Totalizer 1 to n submenu.
process variable isdefined in the Unit
totalizer parameter(→  102) for the
totalizer.

0 kg

Totalizer value–

Displays the current totalizer
counter value.

Signed floating-point
number–Reset all totalizers–

Reset all totalizers to 0 andstart.

• Cancel• Reset + totalize

CancelTotalize

11.6.1

Endress+Hauser

Function scope of "Control Totalizer" parameter

Options

Description

Totalize

The totalizer is started or continues running.

Reset + hold

The totaling process is stopped and the totalizer is reset to 0.

Preset + hold 1)

The totaling process is stopped and the totalizer is set to its defined start value
from the Preset value parameter.

Reset + totalize

The totalizer is reset to 0 and the totaling process is restarted.

125

Operation

Proline Teqwave MW 300 HART

Options

Description

Preset + totalize
Hold
1)

The totalizer is set to the defined start value in the Preset value parameter and thetotaling process is restarted.
Totalizing is stopped.

Visible depending on the order options or device settings

11.6.2

126

1)

Function range of "Reset all totalizers" parameter

Options

Description

CancelNo action is executed and the user exits the parameter.

Reset + totalize

Resets the totalizer to 0 and restarts the totaling process. The previously
aggregated load quantity is thus deleted.

Endress+Hauser

Proline Teqwave MW 300 HART

Operation

11.7

Displaying the measured value history

The Extended HistoROM application package must be enabled in the device (order
option) for the Data logging submenu to appear. This contains all the parameters for themeasured value history.
Data logging is also available via:
• Plant Asset Management Tool FieldCare →  70.
• Web browser
Function range• A total of 1000 measured values can be stored• 4 logging channels• Adjustable logging interval for data logging• Measured value trend for each logging channel displayed in the form of a chart/ ../XXXXXXXX
0.40

%TS

0.20
0.00
- 0.2009 12:40 09 12:41 09 12:42

t
A0053802

 30

Chart of a measured value trend• x-axis: depending on the number of channels selected displays 250 to 1000 measured
values of a process variable.
• y-axis: displays the approximate measured value span and constantly adapts this to theongoing measurement.
If the length of the logging interval or the assignment of the process variables to thechannels is changed, the content of the data logging is deleted.
Navigation"Diagnostics" menu → Data logging‣ Data logging

Endress+Hauser

Assign channel 1→  128

Assign channel 2→  128

Assign channel 3→  128

Assign channel 4→  129

Logging interval→  129

Clear logging data

→  129

Data logging→  129

Logging delay→  129

127

Operation

Proline Teqwave MW 300 HART

Data logging control→  129

Data logging status→  129

Entire logging duration→  129

Parameter overview with brief description
Parameter

Prerequisite

Description

Assign channel 1

The Extended HistoROM
application package isavailable.

Assign process variable tologging channel.

Assign channel 2• The Load rate option is only Assign a process variable toavailable if the volume flow logging channel.
of the medium is read in via
the Current input 1 to n or
the fieldbus.
• The Extended HistoROM
application package isavailable.

Selection / User
entry / Userinterface

Factory setting• Off• Total solids• Temperature• Electronics
temperature• Conductivity• Corrected
conductivity• Load rate *• Current output 1 *• Current output 2 *• Current output 3 *• Current output 4 *Off

For the picklist, see
Assign channel 1
parameter(→  128)

OffFor the picklist, see
Assign channel 1
parameter(→  128)Off

software options
 The
currently enabled are
displayed in the
Software option
overview parameter.
Assign channel 3• The Load rate option is only Assign a process variable toavailable if the volume flow logging channel.
of the medium is read in via
the Current input 1 to n or
the fieldbus.
• The Extended HistoROM
application package isavailable.
software options
 The
currently enabled are
displayed in the
Software option
overview parameter.

128

Endress+Hauser

Proline Teqwave MW 300 HART

Parameter

Assign channel 4

Operation

Prerequisite

Description• The Load rate option is only Assign a process variable toavailable if the volume flow logging channel.
of the medium is read in via
the Current input 1 to n or
the fieldbus.
• The Extended HistoROM
application package isavailable.

Selection / User
entry / Userinterface

Factory setting

For the picklist, see
Assign channel 1
parameter(→  128)Off

software options
 The
currently enabled are
displayed in the
Software option
overview parameter.
Logging interval

The Extended HistoROM
application package isavailable.

Define the logging interval fordata logging. This value
defines the time interval
between the individual data
points in the memory.

0.1 to 3 600.0 s

1.0 s

Clear logging data

The Extended HistoROM
application package isavailable.

Clear the entire logging data.

• Cancel• Clear data

CancelData logging–

Select the type of data logging.

• Overwriting• Not overwriting

Overwriting

Logging delay

In the Data logging
parameter, the Not
overwriting option is selected.

Enter the time delay formeasured value logging.

0 to 999 h

0h

Data logging control

In the Data logging
parameter, the Not
overwriting option is selected.

Start and stop measured value
logging.

• None• Delete + start• StopNone

Data logging status

In the Data logging
parameter, the Not
overwriting option is selected.

Displays the measured value
logging status.

• Done• Delay active• Active• Stopped

Done

Entire logging duration

In the Data logging
parameter, the Not
overwriting option is selected.

Displays the total logging
duration.

Positive floatingpoint number

0s*

Visibility depends on order options or device settings

Endress+Hauser

129

Operation

Proline Teqwave MW 300 HART

11.8

Adjusting the measured value with the aid of wizards

In practice, the measurement typically needs to be adjusted based a reference value (e.g.
lab value) when commissioning the device to ensure optimum measurement performance
during subsequent operation. Repeating this adjustment is recommended if there are
significant changes to the process conditions or following replacement of the sensor
electronics module (ISEM).
The deviation from the value determined by the device can be checked and adjusted if
necessary with the aid of manually taken samples of the medium, which are analyzed in
the laboratory. The lab value is compared with the measured value of the device for thispurpose. The difference between the two values can then be used to decide whether themeasurement performance is sufficient or whether the device should be readjusted basedon the laboratory value.
The device has four wizards to facilitate this process. After starting each wizard, you are
guided through the necessary work steps.
Performing the basic settings for the adjustment:
1.

Commissioning wizard

Adusting the measured value based on the reference value:
2.

1 - Take a sample wizard

3.

2 - Enter lab value wizard

4.

3 - Perform adjustment wizard
Adjustment can be carried out directly via local operation of the device or via the web
server.
The process using the wizards is essentially the same for both operation methods, butmore options and a graphic display are available when operating via the web server inthe 3 - Perform adjustment wizard. Using the web server is therefore recommended.

Online information
Further information on the procedure for performing an adjustment using wizards is alsoavailable online.

11.8.1

Performing the basic settings for the adjustment

The Commissioning wizard is called up via the main menu: Setup → Total solids
adjustment → Commissioning
Commissioning
The Commissioning wizard is used to:• set the system time (when using the wizard for the first time or after disconnecting thedevice from the supply voltage)• set the unit for the solids content for the measured total solids and the lab value• enter the solids density

11.8.2Adjusting the measured value based on the reference value• All three wizards must be run each time the device is adjusted.
• You can call up the wizard options via the operating menu or during local operationfrom the operational display by holding down the Enter key for longer than
3 seconds→  132.

130

Endress+Hauser

Proline Teqwave MW 300 HART

Operation

Taking a sample
The 1 - Take a sample wizard is used to:• set the system time (if not already defined in the Commissioning wizard)• have the total solids measured by the device• calculate the value of the medium sample of the device
The calculated medium sample of the device is the average of the total solids
measured by the device between the start and end of the wizard.
• perform a check as to whether the variability is within the permitted limits You can set
the upper limit in the Maximum variance parameter.
• save the value of the sample calculated by the device (incl. system time and status).
• At the same time as the total solids in the device are measured, a sample must be
taken manually for analysis in the laboratory.
• The installation of sampling points is recommended for taking samples of themedium →  24.
Entering the lab value
The 2 - Enter lab value wizard is used to:• select a unit for entering the sample of the medium taken manually.
• select the sample measured by the device that is to be used for the lab value• enter the lab value of the sample of the medium taken manually• check whether the lab value is within the value range• save the lab value (incl. status and min./max. values if applicable)
Performing the adjustment
The 3 - Perform adjustment wizard is used to:• display the respective medium sample with the corresponding lab values and the numberof adjustments performed.
• select whether a single-point adjustment or a multi-point adjustment is to be performed• In the case of a single-point adjustment, the value of the most recent, valid sample
taken is always selected automatically.
• In the case of a multi-point adjustment, the values of the last ten valid samples arealways selected automatically.
• display the current and new factor and offset• display the current and new value of the total solids• display the time stamp of the completed process and confirm completion.
Each completed adjustment is documented: Diagnostics → Event logbook → Event list
Extended functionality when running the wizard via the web server
In the web server, the samples of the medium are displayed in a table (max. 10 out of 32
stored samples). It is also possible to adjust the selection of samples used for the
adjustment.
• In the case of a single-point adjustment, the value of the most recent, valid sample taken
is always selected automatically. You can determine the most recent sample you want touse by deleting the most recent sample in the table.
• In the case of a multi-point adjustment, the values of the last ten valid samples taken arealways automatically selected. You can determine the samples you want to use for the
adjustment by deleting samples from the table.

Endress+Hauser

131

Operation

Proline Teqwave MW 300 HART

11.8.3Calling up wizards• Commissioning wizard
The wizard is called up via the main menu: Setup → Total solids adjustment→ Commissioning• Take a sample wizard, Enter lab value wizard and Assistent Perform adjustment:
You can call up the wizard options via the operating menu or during local operation fromthe operational display by holding down the Enter key for longer than 3 seconds.
Performing the adjustment process via local operation
1.

Press the Enter key  for longer than 3 seconds.
A0026785

A selection field appears with adjustment options.
2.

Confirm the desired adjustment option in the selection field.
 The available wizards are displayed.

3.

Select the desired wizard and follow the instructions.

Performing the adjustment process via the web server
Access the operating menu via web browser - web server →  60.
1.

After starting the web server, select Main menu > Setup .
Endress+Hauser

Device name
Device tag
Status signal
Total solids adjustment

Measured values Menu Instrument health status Data management Network

Logout (Maintenance)

Main menu > Setup
A0052630

2.

Select Total solids adjustment.
 The available wizards are displayed.

Endress+Hauser

Device name
Device tag
Status signal
Total solids adjustment

Measured values Menu Instrument health status Data management Network

Logout (Maintenance)

Main menu > Setup > Total solids adjustment
1 - Take a sample

2 - Enter lab value

3 - Perform adjustment

A0052631

132

Endress+Hauser

Proline Teqwave MW 300 HART

3.

Operation

Select the desired wizard.
 The individual steps of the wizard are displayed.

Endress+Hauser

Device name
Device tag
Status signal
Total solids adjustment

Measured values Menu Instrument health status Data management Network

Logout (Maintenance)

Main menu > Setup > Total solids adjustment > 1 - Take a sample
Set system time

Start sampling

Sampling started...

ResultFinish

A0053912

4.

Endress+Hauser

Follow the instructions provided by the wizard.
 The wizard guides you through the individual steps.

133Diagnosis and troubleshooting

Proline Teqwave MW 300 HART

12

Diagnosis and troubleshooting

12.1

General troubleshooting

For local display
Fault

Possible causes

Remedial action

Local display is dark, but signal output is within
the valid range

The cable of the display module is not plugged in Insert the plug correctly into the maincorrectly.
electronics module and display module.

Local display dark and no output signals

Supply voltage does not match the voltage
specified on the nameplate.

Apply the correct supply voltage .

Supply voltage has incorrect polarity.

Reverse polarity of supply voltage

No contact between connection cables andterminals

Ensure electrical contact between the cable andthe terminal.

• Terminals are not plugged into the I/O
electronics module correctly.
• Terminals are not plugged into the mainelectronics module correctly.

Check terminals.

• I/O electronics module is defective.
• Main electronics module is defective.

Order spare part →  160.

Local display cannot be read, but signal output is Display is set too bright or too dark.
within the valid range• Set the display brighter by simultaneously
pressing  + .
• Set the display darker by simultaneously
pressing  + .

Local display is dark, but signal output is within
the valid range

Display module is defective.

Order spare part →  160.

Backlighting of local display is red

Diagnostic event with "Alarm" diagnostic
behavior has occurred.

Take remedial measures →  146

Text on local display appears in a language thatcannot be understood.

The selected operating language cannot be
understood.

1. Press  +  for 2 s ("home position").
2. Press .
3. Configure the required language in the
Display language parameter (→  105).

Message on local display:"Communication Error""Check Electronics"

Communication between the display module and • Check the cable and the connector between
the electronics is interrupted.
the main electronics module and display
module.
• Order spare part →  160.

For output signals
Fault

Possible causes

Remedial action

Signal output outside the valid range

Main electronics module is defective.

Order spare part →  160.

Signal output outside the valid current range(< 3.6 mA or > 22 mA)

Main electronics module is defective.
I/O electronics module is defective.

Order spare part →  160.

Device shows correct value on local display, but
signal output is incorrect, though in the valid
range.

Parameter configuration error

Check and adjust parameter configuration.

Device measures incorrectly.

Configuration error or device is operated outside
the application.

1. Check and correct parameter configuration.
2. Observe limit values specified in the"Technical Data".

Unexpected and significant difference withregard to the laboratory value

Formation of buildup on the antennas

1. Remove buildup.
2. Perform a new adjustment →  130.



Buildup usually produces a positive
difference with regard to the laboratory
value.

removing the buildup, make sure
 Whenthat the measuring tube, the antennas andthe temperature sensor are neither
mechanically damaged nor chemicallycorroded.

134

Endress+Hauser

Proline Teqwave MW 300 HART

Diagnosis and troubleshooting

For access
Fault
Write access to parameters is not possible.

Possible causes

Remedial action

Hardware write protection is enabled.

Set the write protection switch on the mainelectronics module to the OFF position→  117.

Current user role has limited accessauthorization.

1. Check user role →  58.
2. Enter correct customer-specific access code→  59.

Missing or incorrectly installed communication
resistor

Install the communication resistor (250 Ω)correctly. Observe the maximum load.

Commubox• Incorrectly connected.
• Incorrectly configured.
• Driver is not installed correctly.
• The USB port on the PC is incorrectly
configured.

Refer to the documentation on Commubox
FXA195 HART:

Unable to connect to web server, FieldCare or
DeviceCare.

WLAN network is not available.

• Check if WLAN reception is present: LED on
display module is lit blue.
• Check if WLAN connection is enabled: LED on
display module flashes blue.
• Switch on instrument function.

Network connection not present or unstable

WLAN network is weak.

• Operating unit outside reception range: Check
network status on operating unit.
• To improve network performance, use an
external WLAN antenna.

Parallel WLAN and Ethernet communication• Check network settings.
• Temporarily enable only the WLAN as aninterface.

Data transfer is active.

Wait until data transfer or current action isfinished.

Connection lost

‣ Check cable connection and power supply.
‣ Refresh the web browser and restart if

Connection via HART protocol is not possible.

Web browser frozen and operation no longer
possible Technical Information TI00404F

necessary.

Display of web browser content is difficult toread or incomplete.

Incomplete or no display of content in the webbrowser

Web browser version used is not optimal.

‣ Use correct web browser version →  60.
‣ Empty the web browser cache.
‣ Restart the web browser.

Unsuitable view settings.

Change the font size/display ratio of the Webbrowser.

• JavaScript is not enabled.
• JavaScript cannot be enabled.

‣ Enable JavaScript.
‣ Enter http://XXX.XXX.X.X.XX/servlet/
basic.html as the IP address.

Operation with FieldCare or DeviceCare via
service interface CDI-RJ45 (port 8000) is notpossible.

Firewall of the PC or network is blocking
communication.

Depending on the settings of the firewall usedon the PC or in the network, the firewall must be
adapted or disabled to allow FieldCare/
DeviceCare access.

Flashing the firmware with FieldCare or
DeviceCare via service interface CDI-RJ45 (port8000 or TFTP ports) is not possible.

Firewall of the PC or network is blocking
communication.

Depending on the settings of the firewall usedon the PC or in the network, the firewall must be
adapted or disabled to allow FieldCare/
DeviceCare access.

Endress+Hauser

135Diagnosis and troubleshooting

Proline Teqwave MW 300 HART

12.2

Diagnostic information via LEDs

12.2.1

Transmitter

Different LEDs in the transmitter provide information on the device status.

1 2 3 4 5A0029629

1
2
3
4
5

Supply voltage
Device status
Not usedCommunication
Service interface (CDI) active

LED

Color/behavior

Meaning

LED off

Supply voltage is off or too low.

Green

Supply voltage is OK.

Off

Firmware error

Green

Device status is OK.

Flashing green

Device is not configured.

Red

A diagnostic event with "Alarm" diagnostic behavior
has occurred.

Flashing red

A diagnostic event with "Warning" diagnostic
behavior has occurred.

Flashing red/green

The device restarts.

Device status (during
start-up)Flashes red slowly

If > 30 seconds: problem with the boot loader.

Flashes red quickly

If > 30 seconds: compatibility problem whenreading the firmware.

3

Not used––4

Communication

LED off

Communication not active.

White

Communication active.

LED off

Not connected or no connection established.

Yellow

Connected and connection established.

Flashing yellow

Service interface active.

1

2

2

5

136

Supply voltage

Device status (normal
operation)

Service interface (CDI)

Endress+Hauser

Proline Teqwave MW 300 HART

Diagnosis and troubleshooting

12.3

Diagnostic information on local display

12.3.1

Diagnostic message

Faults detected by the self-monitoring system of the measuring device are displayed as adiagnostic message in alternation with the operational display.
Operational display in alarm condition

Diagnostic message

2 1

XXXXXXXXX

S

20.50

S801

3
4

Supply voltage

XX

X

S

XXXXXXXXX

i

Menu

5
A0029426-EN

1
2
3
4
5

Status signal
Diagnostic behavior
Diagnostic behavior with diagnostic code
Short text
Operating elements

If two or more diagnostic events are pending simultaneously, only the message of thediagnostic event with the highest priority is shown.
Other diagnostic events that have occurred can be displayed in the Diagnostics menu:• Via parameter →  150• Via submenus →  150
Status signals
The status signals provide information on the state and reliability of the device by
categorizing the cause of the diagnostic information (diagnostic event).
The status signals are categorized according to VDI/VDE 2650 and NAMUR
Recommendation NE 107: F = Failure, C = Function Check, S = Out of Specification, M= Maintenance Required
SymbolMeaning
FailureA device error has occurred. The measured value is no longer valid.
Function check
The device is in service mode (e.g. during a simulation).
Out of specification
The device is being operated:• Outside its technical specification limits (e.g. outside the process temperature range)• Outside the parameter configuration carried out by the user (e.g. full scale value of theconfigured process variable in parameter 20 mA value)
Maintenance required
Maintenance is required. The measured value remains valid.

Endress+Hauser

137

Diagnosis and troubleshooting

Proline Teqwave MW 300 HART

Diagnostic behavior
SymbolMeaning
Alarm• Measurement is interrupted.
• Signal outputs and totalizers assume the defined alarm condition.
• A diagnostic message is generated.
Warning• Measurement is resumed.
• The signal outputs and totalizers are not affected.
• A diagnostic message is generated.

Diagnostic information
The fault can be identified using the diagnostic information. The short text helps you byproviding information about the fault. In addition, the corresponding symbol for the
diagnostic behavior is displayed in front of the diagnostic information on the local display.
Diagnostic information
Diagnostic code
Diagnostic
behavior

Status signal

Diagnostic
numberShort text

↓

↓

↓

441

Curr.output 1

ExampleA0013962

A0013958

NAMUR
NE 1073-digit number

Operating elements
Operating key

MeaningPlus key
In menu, submenu
Opens the message about the remedial measures.
Enter key
In menu, submenu
Opens the operating menu.
If pressed > 3 seconds
Displays the available wizards.

138

Endress+Hauser

Proline Teqwave MW 300 HART

Diagnosis and troubleshooting

12.3.2

Calling up remedial measures

XXXXXXXXX

S

20.50
X

S

XXXXXXXXX

S801
Supply voltage
i

XX

Menu

1.

1

Diagnostic list
Diagnostics 1

S

S801 Supply voltage

Diagnostics 2
Diagnostics 3
2.
2
4
6

Supply voltage(ID:203)
S801 0d00h02m25s
Increase supply voltage

35

3.
A0029431-EN 31

Message for remedial measures

1
2
3
4
5
6

Diagnostic information
Short text
Service ID
Diagnostic behavior with diagnostic code
Operation time when error occurred
Remedial measures

1.

The user is in the diagnostic message.
Press  ( symbol).
 The Diagnostic list submenu opens.

2.

Select the desired diagnostic event with  or  and press  .
 The message about the remedial measures opens.

3.

Press  +  simultaneously.
 The message about the remedial measures closes.

The user is in the Diagnostics menu at an entry for a diagnostics event, e.g. in the
Diagnostic list submenu or Previous diagnostics parameter.

Endress+Hauser

1.

Press .
 The message for the remedial measures for the selected diagnostic event opens.

2.

Press  +  simultaneously.
 The message for the remedial measures closes.

139

Diagnosis and troubleshooting

Proline Teqwave MW 300 HART

12.4

Diagnostic information in the web browser

12.4.1

Diagnostic options

Any faults detected by the measuring device are displayed in the Web browser on thehome page once the user has logged on.
1

2

3
A0031056

1
2
3

Status area with status signal
Diagnostic information
Remedial measures with service ID

In addition, diagnostic events which have occurred can be shown in the Diagnostics
menu:• Via parameter →  150• Via submenu →  150
Status signals
The status signals provide information on the state and reliability of the device by
categorizing the cause of the diagnostic information (diagnostic event).
SymbolMeaning
FailureA device error has occurred. The measured value is no longer valid.
Function check
The device is in service mode (e.g. during a simulation).
Out of specification
The device is being operated:• Outside its technical specification limits (e.g. outside the process temperature range)• Outside the parameter configuration carried out by the user (e.g. full scale value of theconfigured process variable in parameter 20 mA value)
Maintenance required
Maintenance is required. The measured value remains valid.

The status signals are categorized in accordance with VDI/VDE 2650 and NAMUR
Recommendation NE 107.

140

Endress+Hauser

Proline Teqwave MW 300 HART

12.4.2

Diagnosis and troubleshooting

Calling up remedy information

Remedy information is provided for every diagnostic event to ensure that problems can berectified quickly. These measures are displayed in red along with the diagnostic event andthe related diagnostic information.

Endress+Hauser

141

Diagnosis and troubleshooting

Proline Teqwave MW 300 HART

12.5

Diagnostic information in FieldCare or DeviceCare

12.5.1

Diagnostic options

Any faults detected by the measuring device are displayed on the home page of theoperating tool once the connection has been established.
1
Xxxxxx/…/…/
Device name: Xxxxxxx

Mass flow:

12.34

kg/h

Device tag:

Volume flow:

12.34

m³/h

Xxxxxxx

Status signal:

Function check (C)XxxxxxDiagnostics 1:

C485 Simu...

P
P

Remedy information:

Deactivate...

P

Access status tooling:

Mainenance+Operation

+

Setup

+

Diagnostics+

ExpertFailure (F)2Function check (C)Diagnostics 1:

C485 Simulation measured vari...

Remedy information:

Deactivate Simulation (Service...

3

Out of spezification (S)Maintenance required (M)A0021799-EN

1
2
3

Status area with status signal →  137
Diagnostic information →  138
Remedial measures with service ID

In addition, diagnostic events which have occurred can be shown in the Diagnostics
menu:• Via parameter →  150• Via submenu →  150
Diagnostic information
The fault can be identified using the diagnostic information. The short text helps you byproviding information about the fault. In addition, the corresponding symbol for the
diagnostic behavior is displayed in front of the diagnostic information on the local display.
Diagnostic information
Diagnostic code
Diagnostic
behavior

Status signal

Diagnostic
numberShort text

↓

↓

↓

441

Curr.output 1

ExampleA0013962

A0013958

NAMUR
NE 107142

3-digit number

Endress+Hauser

Proline Teqwave MW 300 HART

Diagnosis and troubleshooting

12.5.2

Calling up remedy information

Remedy information is provided for every diagnostic event to ensure that problems can berectified quickly:• On the home page
Remedy information is displayed in a separate field below the diagnostics information.
• In the Diagnostics menu
Remedy information can be called up in the working area of the user interface.
The user is in the Diagnostics menu.

Endress+Hauser

1.

Call up the desired parameter.

2.

On the right in the working area, mouse over the parameter.
 A tool tip with remedy information for the diagnostic event appears.

143Diagnosis and troubleshooting

Proline Teqwave MW 300 HART

12.6

Adapting the diagnostic information

12.6.1

Adapting the diagnostic behavior

Each item of diagnostic information is assigned a specific diagnostic behavior at thefactory. The user can change this assignment for specific diagnostic information in the
Diagnostic behavior submenu.
Expert → System → Diagnostic handling → Diagnostic behavior/ ../Event level
Event no. 044
Warning
Event no. 274
Event no. 801
A0014048-EN

 32

Using the example of the local display

You can assign the following options to the diagnostic number as the diagnostic behavior:
Options

Description

Alarm

The device stops measurement. The signal outputs and totalizers assume the definedalarm condition. A diagnostic message is generated.
The background lighting changes to red.

Warning

The device continues to measure. The signal outputs and totalizers are not affected. A
diagnostic message is generated.

Logbook entry only

The device continues to measure. The diagnostic message is only displayed in the Eventlogbook submenu (Event list submenu) and is not displayed in alternating sequencewith the operational display.

Off

The diagnostic event is ignored, and no diagnostic message is generated or entered.

12.6.2

Adapting the status signal

Each item of diagnostic information is assigned a specific status signal at the factory. The
user can change this assignment for specific diagnostic information in the Diagnostic
event category submenu.
Expert → Communication → Diagnostic event category
Available status signals
Configuration as per HART 7 Specification (Condensed Status), in accordance with
NAMUR NE107.
SymbolMeaning
FailureA device error has occurred. The measured value is no longer valid.
A0013956

Function check
The device is in service mode (e.g. during a simulation).
A0013959

A0013958

144

Out of specification
The device is being operated:• Outside its technical specification limits (e.g. outside the process temperature range)• Outside the parameter configuration carried out by the user (e.g. full scale value of theconfigured process variable in parameter 20 mA value)

Endress+Hauser

Proline Teqwave MW 300 HART

Diagnosis and troubleshooting

SymbolMeaning
Maintenance required
Maintenance is required. The measured value remains valid.
A0013957

Has no effect on the condensed status.
A0023076

Endress+Hauser

145

Diagnosis and troubleshooting

Proline Teqwave MW 300 HART

12.7

Overview of diagnostic information

In the case of some items of diagnostic information, the status signal and the
diagnostic behavior can be changed. Change the diagnostic information →  144
Diagnostic
numberShort text

Remedy instructions

Statussignal[from thefactory]

Diagnostic
behavior[from thefactory]F

Alarm

Diagnostic of sensor
022

Temperature sensor
defective

1. Change main electronic module
2. Change sensor

082

Data storage inconsistent Check module connections

F

Alarm

083

Memory content
inconsistent

1. Restart device
2. Restore S-DAT data
3. Replace S-DAT

F

Alarm

181

Sensor connection faulty

1. Check sensor cable and sensor
2. Perform Heartbeat Verification

F

Alarm

Diagnostic of electronic

146

201

Electronics faulty

1. Restart device
2. Replace electronics

F

Alarm

242

Firmware incompatible

1. Check firmware version
2. Flash or replace electronic module

F

Alarm

252

Module incompatible

1. Check electronic modules
2. Check if correct modules areavailable (e.g. NEx, Ex)
3. Replace electronic modules

F

Alarm

262

Module connection
interrupted

1. Check or replace connection cablebetween sensor electronic module(ISEM) and main electronics
2. Check or replace ISEM or mainelectronics

F

Alarm

270

Main electronics
defective

1. Restart device
2. Replace main electronic module

F

Alarm

271

Main electronics faulty

1. Restart device
2. Replace main electronic module

F

Alarm

272

Main electronics faulty

Restart device

F

Alarm

273

Main electronics
defective

1. Pay attention to display emergency
operation
2. Replace main electronics

F

Alarm

275

I/O module defective

Change I/O module

F

Alarm

276

I/O module faulty

1. Restart device
2. Change I/O module

F

Alarm

283

Memory content
inconsistent

Restart device

F

Alarm

302

Device verification active

Device verification active, please wait.

CWarning 1)

303

I/O 1 to n configuration
changed1. Apply I/O module configuration(parameter "Apply I/O
configuration")
2. Afterwards reload device
description and check wiring

M

Warning311

Sensor electronics(ISEM) faulty

Maintenance required!
Do not reset device

M

Warning

330

Flash file invalid

1. Update firmware of device
2. Restart device

M

WarningEndress+Hauser

Proline Teqwave MW 300 HART

Diagnosis and troubleshooting

Diagnostic
numberShort text

331

Firmware update failed

332

Remedy instructions

Statussignal[from thefactory]

Diagnostic
behavior[from thefactory]

1. Update firmware of device
2. Restart deviceF

Warning

Writing in HistoROM
backup failed

1. Replace user interface board
2. Ex d/XP: replace transmitter

F

Alarm

361

I/O module 1 to n faulty

1. Restart device
2. Check electronic modules
3. Change I/O module or mainelectronics

F

Alarm

372

Sensor electronics(ISEM) faulty

1. Restart device
2. Check if failure recurs
3. Replace sensor electronic module(ISEM)

F

Alarm

373

Sensor electronics(ISEM) faulty

Transfer data or reset device

F

Alarm

375

I/O- 1 to ncommunication failed

1. Restart device
2. Check if failure recurs
3. Replace module rack inclusiveelectronic modules

F

Alarm

378

Supply voltage ISEM
faulty1. If available: Check connection cable F
between sensor and transmitter
2. Replace main electronic module
3. Replace sensor electronic module(ISEM)

Alarm

382

Data storage

1. Insert T-DAT
2. Replace T-DAT

F

Alarm

383

Memory content

Reset device

F

Alarm

387

HistoROM data faulty

Contact service organization

F

Alarm

Diagnostic of configuration

Endress+Hauser

410

Data transfer failed

1. Retry data transfer
2. Check connection

F

Alarm

412

Processing download

Download active, please waitCWarning

431

Trim 1 to n required

Carry out trim

M

Warning437

Configuration
incompatible

1. Update firmware
2. Execute factory reset

F

Alarm

438

Dataset different

1. Check dataset file
2. Check device parameterization
3. Download new device
parameterization

M

Warning

441

Current output 1 to nsaturated

1. Check current output settings
2. Check process

S

Warning 1)

442

Frequency output 1 to nsaturated

1. Check frequency output settings
2. Check process

S

Warning 1)

443

Pulse output 1 to nsaturated

1. Check pulse output settings
2. Check process

S

Warning 1)

444

Current input 1 to nsaturated

1. Check current input settings
2. Check connected device
3. Check process

S

Warning 1)

453

Flow override active

Deactivate flow overrideC

Warning

484

Failure mode simulation
activeDeactivate simulation

C

Alarm

485

Process variable
simulation active

Deactivate simulationC

Warning147

Diagnosis and troubleshooting

Proline Teqwave MW 300 HART

Diagnostic
numberShort text

Remedy instructions

Statussignal[from thefactory]

Diagnostic
behavior[from thefactory]

486

Current input 1 to nsimulation active

Deactivate simulationC

Warning491

Current output 1 to nsimulation active

Deactivate simulationC

Warning492

Frequency output 1 to nsimulation active

Deactivate simulation frequency
outputC

Warning493

Pulse output simulation
activeDeactivate simulation pulse output

C

Warning494

Switch output 1 to nsimulation active

Deactivate simulation switch output

C

Warning495

Diagnostic event
simulation active

Deactivate simulationC

Warning496

Status input 1 to nsimulation active

Deactivate simulation status input

C

Warning520

I/O 1 to n hardware
configuration invalid

1. Check I/O hardware configuration
2. Replace wrong I/O module
3. Plug the module of double pulse
output on correct slot

F

Alarm

537

Configuration

1. Check IP addresses in network
2. Change IP addressF

Warning594

Relay output 1 to nsimulation active

Deactivate simulation switch output

C

WarningDiagnostic of process

148

803

Loop current 1 faulty

1. Check wiring
2. Change I/O module

F

Alarm

832

Electronics temperature
too high

Reduce ambient temperatureS

Warning 1)

833

Electronics temperature
too low

Increase ambient temperatureS

Warning 1)

834

Process temperature too
highReduce process temperatureS

Warning 1)

835

Process temperature too
low

Increase process temperatureS

Warning 1)

844

Total solids range
exceeded

Check range limits

S

Warning 1)

862

Partly filled pipe

1. Verify that the measuring tube isfilled with the medium.
2. Verify that build-up does notinterfere with antennas.

S

Warning 1)

881

Signal to noise ratio too
low

1. Check process conditions
2. Clean measuring tube
3. Replace sensor electronic module(ISEM)

S

Warning882

Input signal faulty

1. Check input signal
parameterization
2. Check external device
3. Check process conditions

F

Alarm

907

Permittivity out ofspecification

Check composition of the medium

S

Warning

908

Volume fraction out ofspecification

1. Perform adjustment
2. Remove build-up on antennas /temperature sensors
3. Check for gas in medium

S

WarningEndress+Hauser

Proline Teqwave MW 300 HART

Diagnosis and troubleshooting

Diagnostic
number1)

Endress+Hauser

Short text

Remedy instructions

Statussignal[from thefactory]

Diagnostic
behavior[from thefactory]909

Conductivity out ofspecification

1. Check process conditions
2. Clean measuring tube
3. Replace sensor electronic module(ISEM)

F

Warning944

Monitoring failed

Check process conditions forHeartbeat Monitoring

S

WarningDiagnostic behavior can be changed.

149

Diagnosis and troubleshooting

Proline Teqwave MW 300 HART

12.8

Pending diagnostic events

The Diagnostics menu allows the user to view the current diagnostic event and theprevious diagnostic event separately.
To call up the measures to rectify a diagnostic event:• Via local display →  139• Via web browser →  141• Via "FieldCare" operating tool →  143• Via "DeviceCare" operating tool →  143
Other pending diagnostic events can be displayed in the Diagnostic list submenu→  150.
Navigation"Diagnostics" menu

 Diagnostics
Actual diagnostics→  150

Previous diagnostics→  150

Operating time from restart→  150

Operating time→  150

Parameter overview with brief description
Parameter
Actual diagnostics

Prerequisite
A diagnostic event has occurred.

Description

User interface

Shows the current occured diagnostic
event along with its diagnostic
information.

Symbol for diagnostic
behavior, diagnostic codeand short message.

two or more messages occur
 Ifsimultaneously,
the message withthe highest priority is shown on
the display.
Previous diagnostics

Two diagnostic events have alreadyoccurred.

Shows the diagnostic event that
occurred prior to the current diagnostic
event along with its diagnostic
information.

Symbol for diagnostic
behavior, diagnostic codeand short message.

Operating time from restart–Shows the time the device has been in
operation since the last device restart.

Days (d), hours (h),minutes (m) and seconds(s)Operating time–

Indicates how long the device has beenin operation.

Days (d), hours (h),minutes (m) and seconds
(s)

12.9

Diagnostics list

Up to 5 currently pending diagnostic events can be displayed in the Diagnostic list
submenu along with the associated diagnostic information. If more than 5 diagnostic
events are pending, the events with the highest priority are shown on the display.
Navigation path
Diagnostics → Diagnostic list
150

Endress+Hauser

Proline Teqwave MW 300 HART

Diagnosis and troubleshooting/ ../Diagnose list
Diagnostics
F273 Main electronic
Diagnostics 2
Diagnostics 3
A0014006-EN

 33

Using the example of the local display

To call up the measures to rectify a diagnostic event:• Via local display →  139• Via web browser →  141• Via "FieldCare" operating tool →  143• Via "DeviceCare" operating tool →  143

Endress+Hauser

151

Diagnosis and troubleshooting

Proline Teqwave MW 300 HART

12.10 Event logbook
12.10.1 Reading out the event logbook
A chronological overview of the event messages that have occurred is provided in the
Events list submenu.
Navigation path
Diagnostics menu → Event logbook submenu → Events list
/ ../Eventlist
F
I1091 Config. change
I1157 Mem.err. ev.list
0d01h19m10s
F311 Electr. failure
A0014008-EN

 34

Using the example of the local display• A maximum of 20 event messages can be displayed in chronological order.
• If the Extended HistoROM application package (order option) is enabled in the device,
the event list can contain up to 100 entries.
The event history includes entries for:• Diagnostic events →  146• Information events →  153
In addition to the operating time when the event occurred, each event is also assigned asymbol that indicates whether the event has occurred or is finished:• Diagnostics event• : Occurrence of the event• : End of the event• Information event
: Occurrence of the event
To call up the measures to rectify a diagnostic event:• Via local display →  139• Via web browser →  141• Via "FieldCare" operating tool →  143• Via "DeviceCare" operating tool →  143
For filtering the displayed event messages →  152

12.10.2 Filtering the event logbook
Using the Filter options parameter you can define which category of event message isdisplayed in the Events list submenu.
Navigation path
Diagnostics → Event logbook → Filter options
Filter categories• All• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• Information (I)152

Endress+Hauser

Proline Teqwave MW 300 HART

Diagnosis and troubleshooting

12.10.3 Overview of information events
Unlike a diagnostic event, an information event is displayed in the event logbook only andnot in the diagnostic list.
Info number

Endress+Hauser

Info name

I1000--------(Device ok)

I1079

Sensor changed

I1089

Power on

I1090

Configuration reset

I1091

Configuration changed

I1092

HistoROM backup deleted

I11362

Total solids measurement adjusted

I1137

Electronics changed

I1151

History reset

I1155

Reset electronics temperature

I1156

Memory error trend

I1157

Memory error event list

I1256

Display: access status changed

I1264

Safety sequence aborted

I1278

I/O module restarted

I1335

Firmware changed

I1361

Web server: login failed

I1397

Fieldbus: access status changed

I1398

CDI: access status changed

I1444

Device verification passed

I1445

Device verification failed

I1450

Monitoring off

I1451

Monitoring on

I1457

Measurement error verification failed

I1459

I/O module verification failed

I1461

Sensor verification failed

I1462

Sensor electronic module verific. failed

I1512

Download started

I1513

Download finished

I1514

Upload started

I1515

Upload finished

I1554

Safety sequence started

I1555

Safety sequence confirmed

I1556

Safety mode off

I1618

I/O module 2 replaced

I1619

I/O module 3 replaced

I1621

I/O module 4 replaced

I1622

Calibration changed

I1624

All totalizers reset

153

Diagnosis and troubleshooting

Proline Teqwave MW 300 HART

Info number

154

Info name

I1625

Write protection activated

I1626

Write protection deactivatedI1627

Web server: login successful

I1628

Display: login successful

I1629

CDI: login successful

I1631

Web server access changed

I1632

Display: login failed

I1633

CDI: login failed

I1634

Reset to factory settings

I1635

Reset to delivery settings

I1639

Max. switch cycles number reached

I1649

Hardware write protection activated

I1650

Hardware write protection deactivated

I1712

New flash file received

I1725

Sensor electronic module (ISEM) changed

I1726

Configuration backup failed

Endress+Hauser

Proline Teqwave MW 300 HART

Diagnosis and troubleshooting

12.11 Resetting the measuring device
The entire device configuration or some of the configuration can be reset to a defined statewith the Device reset parameter (→  112).

12.11.1 Function range of "Device reset" parameter
Options

Description

CancelNo action is executed and the user exits the parameter.

To delivery settings

Every parameter for which a customer-specific default setting was ordered is resetto the customer-specific value. All other parameters are reset to the factory setting.

Restart device

The restart resets every parameter with data stored in volatile memory (RAM) to
the factory setting (e.g. measured value data). The device configuration remains
unchanged.

Restore S-DAT backup

Restores the data that is saved on the S-DAT. Additional information: This function
can be used to resolve the memory issue "083 Memory content inconsistent" or to
restore the S-DAT data when a new S-DAT has been installed.

 This option is displayed only in an alarm condition.

Endress+Hauser

155

Diagnosis and troubleshooting

Proline Teqwave MW 300 HART

12.12 Device information
The Device information submenu contains all parameters that display different
information for device identification.
Navigation"Diagnostics" menu → Device information‣ Device information
Device tag→  156

Serial number→  156

Firmware version→  156

Device name→  157

Manufacturer→  157

Order code→  157

Extended order code 1→  157

Extended order code 2→  157

Extended order code 3→  157

ENP version→  157

Device revision→  157

Device ID→  157

Device type→  157

Manufacturer ID→  157

Parameter overview with brief description
Parameter

Description

User interface

Factory setting

Device tag

Shows name of measuring point.

Max. 32 characters, such as
letters, numbers or special
characters (e.g. @, %, /).

Teqwave M

Serial number

Shows the serial number of the measuring
device.

Max. 11-digit character string
comprising letters andnumbers.

–

Firmware version

Shows the device firmware version installed.

Character string in the format
xx.yy.zz–

156

Endress+Hauser

Proline Teqwave MW 300 HART

Parameter
Device name

Diagnosis and troubleshooting

Description
Shows the name of the transmitter.
name can be found on the The
nameplate of the transmitter.

User interface

Factory setting

Character string comprisingnumbers, letters and special
characters–

Manufacturer

Displays the manufacturer.

Character string comprisingnumbers, letters and special
characters

Endress+Hauser

Order code

Shows the device order code.

Character string composed ofletters, numbers and certain
punctuation marks (e.g. /).

–Extended order code 1

Shows the 1st part of the extended ordercode.

Character string–

Extended order code 2

Shows the 2nd part of the extended ordercode.

Character string–

Character string–

extended order code can also be Thefound on the nameplate of the sensor
and transmitter in the "Ext. ord. cd."
field.
Extended order code 3

Shows the 3rd part of the extended order
code.
extended order code can also be Thefound on the nameplate of the sensor
and transmitter in the "Ext. ord. cd."
field.

ENP version

Shows the version of the electronic
nameplate (ENP).

Character string

2.02.00

Device revision

Displays the device revision with which thedevice is registered with the HART
FieldComm Group.

2-digit hexadecimal number1

Device ID

Shows the device ID for identifying thedevice in a HART network.

6-digit hexadecimal number–

Device type

Displays the device type with which thedevice is registered with the HART
FieldComm Group.

Hexadecimal number

11B3

Manufacturer ID

Displays the device's manufacturer ID
2-digit hexadecimal number
registered with the HART FieldComm Group.

Endress+Hauser

0x11 (for Endress+Hauser)

157

Diagnosis and troubleshooting

Proline Teqwave MW 300 HART

12.13 Firmware history
Releasedate

03.2024

Firmwareversion01.00.zz

Order codefor"Firmwareversion"Option 76

Firmware
Changes

Original firmware

Documentation typeOperating
instructions

Documentation

BA02320D/06/EN/03.24

It is possible to flash the firmware to the current version or the previous version using
the service interface.
For the compatibility of the firmware version with the previous version, the installed
device description files and operating tools, observe the information about the device
in the "Manufacturer's information" document.
The manufacturer's information is available:• In the Download Area of the Endress+Hauser web site: www.endress.com →Downloads• Specify the following details:• Product root: e.g. 4W3B
The product root is the first part of the order code: see the nameplate on thedevice.
• Text search: Manufacturer's information• Media type: Documentation – Technical Documentation

158

Endress+Hauser

Proline Teqwave MW 300 HART

Maintenance

13

Maintenance

13.1

Maintenance work

No special maintenance work is required.

13.1.1

Exterior cleaning

When cleaning the exterior of measuring devices, always use cleaning agents that do notattack the surface of the housing or the gaskets.

13.2

Endress+Hauser services

Endress+Hauser offers a wide variety of services for maintenance such as function checks
at the place of manufacture, maintenance service or device tests.
Your Endress+Hauser Sales Center can provide detailed information on the services.

Endress+Hauser

159

Repair

Proline Teqwave MW 300 HART

14

Repair

14.1

General notes

14.1.1

Repair and conversion concept

The Endress+Hauser repair and conversion concept provides for the following:• The measuring devices have a modular design.
• Spare parts are grouped into logical kits with the associated Installation Instructions.
• Repairs are carried out by Endress+Hauser Service or by appropriately trained customers.
• Certified devices can only be converted to other certified devices by Endress+Hauser
Service or at the factory.

14.1.2

Notes for repair and conversion

For repair and conversion of a measuring device, observe the following notes:‣ Use only original Endress+Hauser spare parts.
‣ Carry out the repair according to the Installation Instructions.
‣ Observe the applicable standards, federal/national regulations, Ex documentation (XA)
and certificates.
‣ Document all repairs and conversions and enter the details in Netilion Analytics.

14.2

Spare parts

Device Viewer (www.endress.com/deviceviewer):
All the spare parts for the measuring device, along with the order code, are listed here andcan be ordered. If available, users can also download the associated Installation
Instructions.
Measuring device serial number:• Is located on the nameplate of the device.
• Can be read out via the Serial number parameter (→  156) in the Device
information submenu.

14.3

Endress+Hauser services

Endress+Hauser offers a wide range of services.
Your Endress+Hauser Sales Center can provide detailed information on the services.

14.4

ReturnThe requirements for safe device return can vary depending on the device type andnational legislation.

160

1.

Refer to the web page for information:
https://www.endress.com/support/return-material
 Select the region.

2.

If returning the device, pack the device in such a way that it is reliably protectedagainst impact and external influences. The original packaging offers the bestprotection.

Endress+Hauser

Proline Teqwave MW 300 HART

Repair

14.5

Disposal

If required by the Directive 2012/19/EU on waste electrical and electronic equipment(WEEE), the product is marked with the depicted symbol in order to minimize thedisposal of WEEE as unsorted municipal waste. Do not dispose of products bearing thismarking as unsorted municipal waste. Instead, return them to the manufacturer for
disposal under the applicable conditions.

14.5.1
1.

Removing the measuring device

Switch off the device.

LWARNING
Danger to persons from process conditions!
‣ Beware of hazardous process conditions such as pressure in the measuring device, hightemperatures or aggressive media.
2.

Carry out the mounting and connection steps from the "Mounting the measuring
device" and "Connecting the measuring device" sections in reverse order. Observe thesafety instructions.

14.5.2

Disposing of the measuring device

LWARNING
Danger to personnel and environment from fluids that are hazardous to health.
‣ Ensure that the measuring device and all cavities are free of fluid residues that arehazardous to health or the environment, e.g. substances that have permeated into
crevices or diffused through plastic.
Observe the following notes during disposal:‣ Observe valid federal/national regulations.
‣ Ensure proper separation and reuse of the device components.

Endress+Hauser

161

Accessories

Proline Teqwave MW 300 HART

15

Accessories

Various accessories, which can be ordered with the device or subsequently from Endress+Hauser, are available for the device. Detailed information on the order code in question isavailable from your local Endress+Hauser sales center or on the product page of theEndress+Hauser website: www.endress.com.

15.1

Device-specific accessories

15.1.1For the transmitter

Accessories

Description

Proline 300 transmitter

Transmitter for replacement Use the order code to define the following
specifications:• Approvals• Output• Input• Display/operation• Housing• Software Order code: 4X3BXX
 Installation instructions EA01xxxD
Remote display andoperating module
DKX001

Remote display and operating module DKX001
If ordered directly with the device:
Order code for "Display; operation", option O "Remote display 4-line, illuminated;
10 m (30 ft) cable; touch control"
If the device is ordered separately:
Order code for "Display; operation", option M "None, prepared for remote display"
If the remote display and operating module is ordered separately:
Via the separate product structure DKX001
Mounting bracket for DKX001• If ordered directly: order code for "Accessory enclosed", option RA "Mounting
bracket, pipe 1/2""• If ordered subsequently: order number: 71340960
Optional connecting cable available
Cable lengths available for order:• If the remote display and operating module DKX001 is ordered together with thedevice: 10 m (35 ft)• If the remote display and operating module is ordered separately: order code
DKX001, option:• A: 5 m (15 ft)• B: 10 m (35 ft)• D: 20 m (30 ft)• E: 30 m (100 ft)
futher information on display and operating module DKX001, see Special
 For
Documentation SD01763D. →  192

162

Endress+Hauser

Proline Teqwave MW 300 HART

Accessories

External WLAN antenna

External WLAN antenna with 1.5 m (59.1 in) connecting cable and two angle
brackets. Order code for "Accessory enclosed", option P8 "Wireless antenna widearea". Further information on the WLAN interface →  68

 Order number: 71351317
 Installation Instructions EA01238D
Weather protection cover

Is used to protect the measuring device from the effects of the weather: e.g.
rainwater, excess heating from direct sunlight.

 Order number: 71343505
 Installation Instructions EA01160D
15.1.2For the sensor

Accessories

Description

Mounting kit

Consists of:• Screws/mounting bolts• Gaskets• Washers• Nuts Order number: DK4M
15.2

Communication-specific accessories

Accessories

Description

Commubox FXA195
HART

For intrinsically safe HART communication with FieldCare via the USB port.

HART Loop Converter
HMX50

 Technical Information TI00404F
Is used to evaluate and convert dynamic HART process variables to analog current
signals or limit values.
Information TI00429F
 •• Technical
Operating Instructions BA00371F

Fieldgate FXA42

Transmission of the measured values of connected 4 to 20 mA analog measuring
devices, as well as digital measuring devices
Technical Information TI01297S
 •• Operating
Instructions BA01778S• Product page: www.endress.com/fxa42

Field Xpert SMT50

The Field Xpert SMT70 tablet PC for device configuration enables mobile plant
asset management in non-hazardous areas. It is suitable for commissioning andmaintenance staff to manage field instruments with a digital communicationinterface and to record progress.
This tablet PC is designed as an all-in-one solution with a preinstalled driver libraryand is an easy-to-use, touch-sensitive tool which can be used to manage field
instruments throughout their entire life cycle.
Technical Information TI01342S
 •• Operating
Instructions BA01709S• Product page: www.endress.com/smt50

Endress+Hauser

163

Accessories

Proline Teqwave MW 300 HART

Field Xpert SMT70

The Field Xpert SMT70 tablet PC for device configuration enables mobile plant
asset management in hazardous and non-hazardous areas. It is suitable forcommissioning and maintenance staff to manage field instruments with a digitalcommunication interface and to record progress.
This tablet PC is designed as an all-in-one solution with a preinstalled driver libraryand is an easy-to-use, touch-sensitive tool which can be used to manage field
instruments throughout their entire life cycle.
Technical Information TI01342S
 •• Operating
Instructions BA01709S• Product page: www.endress.com/smt70

Field Xpert SMT77

The Field Xpert SMT77 tablet PC for device configuration enables mobile plant
asset management in areas categorized as Ex Zone 1.
Technical Information TI01418S
 •• Operating
Instructions BA01923S• Product page: www.endress.com/smt77

15.3

Service-specific accessories

Accessories

Description

Applicator

Software for selecting and sizing Endress+Hauser measuring devices:• Choice of measuring devices for industrial requirements• Graphic display of the calculation results• Determination of the partial order code, administration, documentation andaccess to all project-related data and parameters over the entire life cycle ofa project.
Applicator is available:
Via the Internet: https://portal.endress.com/webapp/applicator

Netilion

lloT ecosystem: Unlock knowledge
With the Netilion lloT ecosystem, Endress+Hauser enables you to optimize
your plant's performance, digitize workflows, share knowledge and improve
collaboration.
Drawing on decades of experience in process automation, Endress+Hauser
offers process industries an lloT ecosystem that provides customers with datadriven insights. These insights can be used to optimize processes, thus leading
to higher plant availability, efficiency and reliability - and ultimately togreater profitability.
www.netilion.endress.com

FieldCare

FDT-based plant asset management tool from Endress+Hauser.
It can configure all smart field units in your system and helps you manage
them. By using the status information, it is also a simple but effective way ofchecking their status and condition.

 Operating Instructions BA00027S and BA00059S
DeviceCare

Tool to connect and configure Endress+Hauser field devices.

 Innovation brochure IN01047S
Retrofit kit for display/WLAN

Retrofitting the device with a display with WLAN
The retrofit kit contains all the necessary parts.
number: DKZ001
 •• Order
You must state the serial number of the device to be converted whenplacing the order.

Retrofit kit for inputs/outputs• For subsequent switching of the functionality of inputs/outputs 2 and 3
using a serial number-based license code
• For subsequent hardware expansion of empty slots for inputs/outputs 2
and 3 using a serial number-based license code and hardware

 Order number: DKZ004

164

Endress+Hauser

Proline Teqwave MW 300 HART

15.4
Accessories

Accessories

System components
Description

Proline flowmeter Promag To calculate the load rate, you need to know the volume flow of the medium. You
400
can measure this value using a flowmeter, e.g. the Proline Promag W 400.
The measured value can be read in as an input signal via the HART protocol or via
the 4 to 20 mA current input from the Teqwave MW and used to calculate the loadrate.
The calculated load rate can be shown on the local display and output as an output
signal.

 Technical Information Proline Promag W 400: TI01046D
 Order number Proline Promag W 400: 5W4C**-

Endress+Hauser

165

Technical data

Proline Teqwave MW 300 HART

16

Technical data

16.1

Application

The measuring device is suitable only for the measurement of solids in water-based
liquids.
Depending on the version ordered, the measuring device can also measure potentiallyexplosive media.
To ensure that the device remains in proper operating condition for its service life, use themeasuring device only for media against which the process-wetted materials are
sufficiently resistant.

16.2
Measuring principle

Function and system design

Microwave transmission
Total solids measurement via microwave transmission: The device measures the time offlight and absorption of microwave transmission between two antennas facing each otherin the measuring tube. Based on these variables, the permittivity of the medium can becalculated, for example.
The temperature of the medium is determined in order to calculate the compensation
factor due to temperature effects. This signal corresponds to the process temperature andis also available as an output signal.
The conductivity of the medium is derived from the change in amplitude and phase of the
microwave signal.

Measuring system

The device consists of a transmitter and a sensor.
The device is available as a compact version:
The transmitter and sensor form a mechanical unit.
Information on the structure of the device →  13

16.3
Measured variable

Input

Direct measured variables• Total solids• Electrical conductivity• Medium temperature
Calculated measured variables
Load rate
The load rate can only be calculated with the volume flow of the medium. This measured
value must be read in via a flowmeter→  167.
Example of calculation:• Volume flow read in by flowmeter: 100 l/min• Total solids measured by Teqwave MW 300 : 10 g/l
Calculated load rate: 1 kg/min

166

Endress+Hauser

Proline Teqwave MW 300 HART

Measuring range

Technical data

Total solids
0 to 500 g/l (0 to 31 lb/ft3), 0 to 50 %TS
Medium temperature
0 to 80 °C (32 to 176 °F)
Electrical conductivity
To ensure correct measurement, the electrical conductivity of the medium must notexceed the measuring range of the temperature-compensated electrical conductivity.
Measuring range for temperature-compensated electrical conductivity at 25 °C (77 °F)
Nominal diameter

Input signal

Electrical conductivity

[mm]

[in]

[mS/cm]50

20 to 10080

30 to 85100

40 to 50150

60 to 20200

80 to 14.5250

100 to 14.5300

120 to 14.5

External measured values
To calculate the load rate, you need to know the volume flow of the medium. You canmeasure this value using a flowmeter, e.g. the Proline Promag W 400.
The volume flow can be read in as an input signal via the HART protocol or via the 4 to 20
mA current input from the Teqwave MW and used to calculate the load rate.
The Proline W Promag 400 flowmeter can be ordered from Endress+Hauser→  165.
Current input
Measured variables can be transferred from the automation system to the device via thecurrent input →  167.
Digital communication
Measured variables can be transferred from the automation system to the device via the
HART protocol.
Current input 4 to 20 mA

Endress+Hauser

Order code

"Output; input 2" (021) or "Output; input 3" (022):
Option I: 4 to 20 mA input

Current input

0/4 to 20 mA (active/passive)

Current range• 4 to 20 mA (active)• 0/4 to 20 mA (passive)

Resolution

1 µA

Voltage drop

Typically: 0.6 to 2 V for 3.6 to 22 mA (passive)

Maximum input voltage≤ 30 V (passive)

167

Technical data

Proline Teqwave MW 300 HART

Open-circuit voltage≤ 28.8 V (active)

Possible input variables

Volume flow of the medium for calculating the load rate

Status input

168

Order code

"Output; input 2" (021) or "Output; input 3" (022):
Option J: status input

Maximum input values• DC –3 to 30 V
• If status input is active (ON): Ri >3 kΩ

Response time

Configurable: 5 to 200 ms

Input signal level• Low signal (low): DC –3 to +5 V
• High signal (high): DC 12 to 30 V

Assignable functions• Disable• Flow override• Reset totalizer (load rate)

Endress+Hauser

Proline Teqwave MW 300 HART

16.4
Output signal

Technical data

OutputCurrent output 4 to 20 mA HART
Order code

"Output; input 1" (020):
Option BA: current output 4 to 20 mA HART

Signal mode

Can be set to:• Active• Passive

Current range

Can be set to:• 4 to 20 mA NAMUR• 4 to 20 mA US• 4 to 20 mA• 0 to 20 mA (only if the signal mode is active)• Fixed current

Open-circuit voltage

DC 28.8 V (active)

Maximum input voltage

DC 30 V (passive)

Load

250 to 700 Ω

Resolution

0.38 µA

Damping

Configurable: 0 to 999.9 s

Assignable
process variables• Total solids• Conductivity• Temperature• Electronics temperature• Totalizer (load rate)• Load rate

Current output 4 to 20 mA

Endress+Hauser

Order code

"Output; input 2" (021) or "Output; input 3" (022):
Option B: current output 4 to 20 mA

Signal mode

Can be set to:• Active• Passive

Current range

Can be set to:• 4 to 20 mA NAMUR• 4 to 20 mA US• 4 to 20 mA• 0 to 20 mA (only if the signal mode is active)• Fixed current

Maximum output values

22.5 mA

Open-circuit voltage

DC 28.8 V (active)

Maximum input voltage

DC 30 V (passive)

Load

0 to 700 Ω

Resolution

0.38 µA

Damping

Configurable: 0 to 999.9 s

Assignable
process variables• Total solids• Conductivity• Temperature• Electronics temperature• Load rate169

Technical data

Proline Teqwave MW 300 HART

Pulse/frequency/switch output
Order code

"Output; input 2" (021) or "Output; input 3" (022):
Option E: Pulse/frequency/switch output

FunctionCan be configured as pulse, frequency or switch output

VersionOpen collector
Can be set to:• Active• Passive• Passive NAMUR

Maximum input values

DC 30 V, 250 mA (passive)

Open-circuit voltage

DC 28.8 V (active)

Voltage drop

For 22.5 mA: ≤ DC 2 V

Pulse output
Maximum input values

DC 30 V, 250 mA (passive)

Maximum output current

22.5 mA (active)

Open-circuit voltage

DC 28.8 V (active)

Pulse width

Configurable: 0.05 to 2 000 ms

Maximum pulse rate

10 000 Impulse/s

Pulse value

Configurable

Assignable
process variables

Totalizer (load rate)

Frequency output
Maximum input values

DC 30 V, 250 mA (passive)

Maximum output current

22.5 mA (active)

Open-circuit voltage

DC 28.8 V (active)

Output frequency

Configurable: end value frequency 2 to 10 000 Hz(f max = 12 500 Hz)

Damping

Configurable: 0 to 999.9 s

Pulse/pause ratio

1:1

Assignable
process variables• Total solids• Conductivity• Temperature• Electronics temperature• Load rate

Switch output

170

Maximum input values

DC 30 V, 250 mA (passive)

Open-circuit voltage

DC 28.8 V (active)

Switching behavior

Binary, conductive or non-conductive

Switching delay

Configurable: 0 to 100 s

Number of switchingcyclesUnlimited

Assignable functions• Disable• On• Diagnostic behavior• Limit value:• Total solids• Conductivity• Temperature• Electronics temperature• Partially filled pipe
• Totalizer (load rate)• Load rate

Endress+Hauser

Proline Teqwave MW 300 HART

Technical data

Relay output
Order code

"Output; input 2" (021) or "Output; input 3" (022):
Option H: relay output

Function

Switch output

Version

Relay output, galvanically isolated

Switching behavior

Can be set to:• NO (normally open), factory setting• NC (normally closed)

Maximum switchingcapacity (passive)• DC 30 V, 0.1 A• AC 30 V, 0.5 A

Assignable functions

• Off
• On
• Diagnostic behavior• Limit value:• Total solids• Conductivity• Temperature• Partially filled pipe
• Totalizer (load rate)• Load rate

User-configurable input/output

Signal on alarm

Order code

"Output; input 2" (021) or "Output; input 3" (022):
Option D: user-configurable input/output

Function

One specific input or output can be assigned to the user-configurable input/
output (configurable I/O) when commissioning the device.

Possible assignment• Current output 4 to 20 mA• Pulse/frequency/switch output• Current input 0/4 to 20 mA• Status input

Technical values of the
inputs and outputs

Correspond to the inputs and outputs described in this section

HART current output
Device diagnostics

Device condition can be read out via HART Command 48

Current output 0/4 to 20 mA
4 to 20 mA
Failure mode

Choose from:• 4 to 20 mA in accordance with NAMUR recommendation NE 43• 4 to 20 mA in accordance with US• Min. value: 3.59 mA• Max. value: 22.5 mA• Definable value between: 3.59 to 22.5 mA• Actual value• Last valid value

0 to 20 mA
Failure modeEndress+Hauser

Choose from:• Maximum alarm: 22 mA• Definable value between: 0 to 20.5 mA

171

Technical data

Proline Teqwave MW 300 HART

Pulse/frequency/switch output
Pulse output
Fault mode

Choose from:• Actual value• No pulses

Frequency output
Fault mode

Choose from:• Actual value• 0 Hz• Definable value between: 2 to 12 500 Hz

Switch output
Fault mode

Choose from:• Current status• Open• Closed

Relay output
Failure mode

Choose from:• Current status• Open• Closed

Local display
Plain text display

With information on cause and remedial measures

Backlight

Red lighting indicates a device error.

Status signal as per NAMUR recommendation NE 107
Interface/protocol• Via digital communication:
HART protocol• Via service interface• CDI-RJ45 service interface• WLAN interface
Plain text display

With information on cause and remedial measures

Web browser
Plain text display

With information on cause and remedial measures

Light emitting diodes (LED)
Status information

Status indicated by various light emitting diodes
The following information is displayed depending on the device version:• Supply voltage active• Data transmission active• Device alarm/error has occurred Diagnostic information via light emitting diodes →  136
172

Endress+Hauser

Proline Teqwave MW 300 HART

Technical data

Load

Output signal →  169

Ex connection data

Safety-related values
Order code for "Output; input 1"
OptionOutput/input type

Safety-related values for output/input 1
26 (+)BA

Current output 4 to 20 mA HART

27 (–)UN = 30 VDC
UM = 250 VAC

Order code for "Output; input 2" and "Output; input 3"
OptionOutput/input type

Safety-related values for output/input
2
24 (+)B

Current output 4 to 20 mA

UN = 30 VDC
UM = 250 VAC

D

User-configurable input/output

UN = 30 VDC
UM = 250 VAC

E

Pulse/frequency/switch output

UN = 30 VDC
UM = 250 VAC

H

Relay output

I

Current input 0/4 to 20 mA

UN = 30 VDC
UM = 250 VAC

J

Status input

UN = 30 VDC
UM = 250 VAC

3
25 (–)

22 (+)

23 (–)UN = 30 VDC
IN =100 mADC/500 mAAC
UM = 250 VAC

Galvanic isolation

The outputs are galvanically isolated from one another and from earth (PE).

Protocol-specific data

Protocol-specific data
Manufacturer ID

0x11

Device type ID

11B3

HART protocol revision7

Device description files(DTM, DD)

Information and files available at:www.endress.com

HART load

Min. 250 Ω

System integration

Information regarding system integration →  73.
• Measured variables via HART protocol• Burst Mode functionality

16.5
Terminal assignment
Available device plugs
Endress+Hauser

Power supply→  35
Device plugs may not be used in hazardous areas!
173

Technical data

Proline Teqwave MW 300 HART

Device plug for connecting to the service interface:
Order code for "Accessory mounted"
Option NB, RJ45 M12 adapter (service interface) →  175
Order code for "Accessory mounted", option NB "Adapter RJ45 M12 (service
interface)"
Order code

Supply voltage

Cable entry/coupling"Accessory mounted"

Cable entry
2

Cable entry
3

NB

Plug M12 × 1–

Order code
"Power supply"

Terminal voltage
DC 24 V

Option I

Power consumption

Frequency range±20%

AC 100 to 240 V –15…+10%–
50/60 Hz

TransmitterMax. 10 W (active power)switch-on current

Current consumption

Max. 36 A (<5 ms) as per NAMUR Recommendation NE 21

Transmitter• Max. 400 mA (24 V)• Max. 200 mA (110 V, 50/60 Hz; 230 V, 50/60 Hz)

Power supply failure• Totalizer stops at the last value measured.
• Depending on the device version, the configuration is retained in the device memory orin the plug-in memory (HistoROM DAT).
• Error messages (incl. total operated hours) are stored.

Overcurrent protection
element

The device must be operated with a dedicated circuit breaker, as it does not have an
ON/OFF switch of its own.
• The circuit breaker must be easy to reach and labeled accordingly.
• Permitted nominal current of the circuit breaker: 2 A up to maximum 10 A.

Electrical connection→  33

Potential equalization

Requirements• Pay attention to in-house grounding concepts• Take account of operating conditions like the pipe material and grounding• Connect the medium and transmitter to the same electrical potential.
• Use a ground cable with a minimum cross-section of 6 mm2 (0.0093 in2) and a cable lug
for potential equalization connections

Terminals

174

Spring-loaded terminals: Suitable for strands and strands with ferrules.
Conductor cross-section 0.2 to 2.5 mm2 (24 to 12 AWG).

Endress+Hauser

Proline Teqwave MW 300 HART

Cable entries

Technical data

• Cable gland: M20 × 1.5 with cable ⌀ 6 to 12 mm (0.24 to 0.47 in)• Thread for cable entry:• NPT ½"
• G ½"• M20
Optional: M12 device plug for connection to the service interface
Order code for "Accessories mounted", option NB: "Adapter RJ45 M12 (service
interface)"→  175

Pin assignment, device plug

Service interface for pin assignment, device plug
Order code for "Accessories mounted", option NB: Adapter RJ45 M12 (service interface)
Pin

2

3

1

4
A0032047

Assignment

1

+

Tx

2

+

Rx

3

-

Tx

4

-

Rx

CodingPlug/socket

D

SocketCable specification→  33

Overvoltage protection

Mains voltage fluctuations→  174

Overvoltage category

Overvoltage category II

Short-term, temporary overvoltage

Between cable and ground up to 1200 V, for max. 5 s

Long-term, temporary overvoltage

Between cable and ground up to 500 V

16.6
Accuracy of outputs

Performance characteristics

The outputs have the following base accuracy specifications.
Current output
Accuracy±5 µA

Pulse/frequency output
Accuracy

Endress+Hauser

Max. ±50 ppm of the measured value (over the entire ambient temperature
range)

175

Technical data

Repeatability

Proline Teqwave MW 300 HART

Total solids
Nominal diameterStandard deviation of total solids

[mm]

[in]

[%TS]

50 to 80

2 to 30.02

100 to 300

4 to 12

0.01

Medium temperature± 0.5 °C (± 0.9 °F)
Electrical conductivity
[%]
20
18
16
14
12
10
8
6
4
2
0
0

2000

4000

6000

8000

[μS/cm]
10000
A0052544

 35

Influence of ambient
temperature

Repeatability in % of measured value - electrical conductivity [μS/cm]

Current output
Temperature coefficientMax. 1 μA/°C

Pulse/frequency output
Temperature coefficient

16.7
Mounting requirements

Mounting→  21

16.8
Ambient temperature
range

No additional effect.

Environment

Transmitter and sensor–20 to +60 °C (–4 to +140 °F)
Readability of the display may be impaired at temperatures outside the temperature
range.

176

Endress+Hauser

Proline Teqwave MW 300 HART

Technical data

If operating the device outdoors:• Install the measuring device in a shady location.
• Avoid direct sunlight, particularly in warm climatic regions.
• Avoid direct exposure to weather conditions.
• Protect the display against impact.
• Protect the display from abrasion, e.g. caused by sand in desert areas.
A weather protection cover is available as an accessory.→  162

Storage temperature–20 to +60 °C (–4 to +140 °F)• Protect the measuring device against direct sunlight during storage in order to avoid
unacceptably high surface temperatures.
• Select a storage location where moisture cannot collect in the measuring device.

Relative humidity

The device is suitable for use in outdoor and indoor areas with a relative humidity of
4 to 95%.

Operating height

According to EN 61010-1• ≤ 2 000 m (6 562 ft)• > 2 000 m (6 562 ft) with additional overvoltage protection (e.g. Endress+Hauser HAWSeries)Degree of protection

Measuring device• IP66/67, Type 4X enclosure, suitable for pollution degree 4• When the housing is open: IP20, Type 1 enclosure, suitable for pollution degree 2• Display module: IP20, Type 1 enclosure, suitable for pollution degree 2External WLAN antenna
IP67

Vibration and shock
resistance• Sinusoidal vibration according to IEC 60068-2-6• 2 to 8.4 Hz, 3.5 mm peak
• 8.4 to 2 000 Hz, 1 g peak
• Broadband random vibration according to IEC 60068-2-64• 10 to 200 Hz, 0.003 g2/Hz• 200 to 2 000 Hz, 0.001 g2/Hz• Total: 1.54 g rms• Half-sine shocks according to IEC 60068-2-27
6 ms 30 g• Rough handling shocks according to IEC 60068-2-31

Mechanical load

Transmitter housing:• Protect against mechanical effects, such as shock or impact.
• Do not use as a ladder or climbing aid.

Electromagnetic
compatibility (EMC)

As per IEC/EN 61326

Endress+Hauser

177

Technical data

Proline Teqwave MW 300 HART

16.9
Medium temperature range
Electrical conductivity

Process

0 to +80 °C (+32 to +176 °F)
To ensure correct measurement, the electrical conductivity of the medium must notexceed the measuring range of the temperature-compensated electrical conductivity.
Measuring range for temperature-compensated electrical conductivity at 25 °C (77 °F)
Nominal diameter

Pressure/temperature
ratings

Electrical conductivity

[mm]

[in]

[mS/cm]50

20 to 10080

30 to 85100

40 to 50150

60 to 20200

80 to 14.5250

100 to 14.5300

120 to 14.5

For an overview of the pressure-temperature ratings for the process connections, see
the Technical Information.→  191

Flow velocity

If there is a risk of deposits building up in the measuring tube, as a result of grease forexample, a flow velocity of >2 m/s (6.5 ft/s) is recommended.

Thermal insulation→  25

Static pressure≥ 1.5 bar (21.8 psi), to avoid outgassing of the medium
Installation near pumps →  22

Vibrations

Information on the measuring system's resistance to vibration and shock →  177

16.10 Mechanical construction
Design, dimensions

Weight

For the dimensions and installation lengths of the device, see the "Technical
Information" document, "Mechanical construction" section .→  191
All values: weight without packaging material
DeviceNominal diameter

178

Weight

[mm]

[in]

50

2

10.6 kg (23.4 lb)

80

3

10.9 kg (24.0 lb)

Endress+Hauser

Proline Teqwave MW 300 HART

Technical data

Nominal diameter

Materials

Weight

[mm]

[in]

100

5

12.6 kg (27.7 lb)

150

617.1 kg (37.8 lb)

200

823.9 kg (52.7 lb)

250

10

32.8 kg (72.3 lb)

300

12

37.8 kg (83.4 lb)

Transmitter
Housing
Order code for "Transmitter housing":
Option A "Aluminum coated": aluminum, AlSi10Mg, coated
Window material
Order code for "Transmitter housing":
Option A "Aluminum, coated": glass
Cable entries/cable glands

1

2
3
4
A0028352

 36
1
2
3
4

Possible cable entries/cable glands

Female thread M20 × 1.5
Cable gland M20 × 1.5
Adapter for cable entry with female thread G ½" or NPT ½"
Device plug

Cable entry/cable glandMaterial

Compression fitting M20 × 1.5

Order code for "Housing", option A"Aluminum, coated":• Non-Ex: plastic• Z2, D2, Ex d/de: brass with plastic
Order code for "Housing", option L "Casting,stainless":
Stainless steel, 1.4404 (316L)

Adapter for cable entry with female thread G ½"

Nickel-plated brass

Adapter for cable entry with female thread NPT ½"

Endress+Hauser

179

Technical data

Proline Teqwave MW 300 HART

Device plug
Electrical connection

Material

Plug M12x1• Socket: Stainless steel, 1.4404 (316L)• Contact housing: Polyamide• Contacts: Gold-plated brass

Measuring pipe
Stainless steel: 1.4408 as per DIN EN 10213 (CF3M as per ASME A351)
Antennas• Parts in contact with medium: ceramic• Antenna bracket: stainless steel: 1.4435 (316L)
Temperature sensor
Stainless steel: 1.4435 (316L)
Accessories
Protective cover
Stainless steel, 1.4404 (316L)
External WLAN antenna• Antenna: ASA plastic (acrylonitrile styrene acrylate) and nickel-plated brass• Adapter: Stainless steel and nickel-plated brass• Cable: Polyethylene• Plug: Nickel-plated brass• Angle bracket: Stainless steel
Mounting kit
For installing the sensor• Screws/mounting bolts, nuts and washers: stainless steel, 1.4301/304, 1.4306/1.4307• Gaskets: aramid fibers, with NBR binder
Remote display and operating module DKX001
Housing material: AlSi10Mg, coated

16.11 Display and user interface
Operation concept

Operator-oriented menu structure for user-specific tasks• Commissioning• Operation• Diagnosis• Expert level
Quick and safe commissioning• Guided menus ("Make-it-run" wizards) for applications• Menu guidance with brief descriptions of the individual parameter functions• Access to the device via web server• WLAN access to the device via mobile handheld terminal, tablet or smart phone

180

Endress+Hauser

Proline Teqwave MW 300 HART

Technical data

Reliable operation• Operation in local language• Uniform operating philosophy applied to device and operating tools• Guided menus (wizards) for adjusting the device using medium samples• If replacing electronic modules, transfer the device configuration via the integrated
memory (HistoROM backup) which contains the process and measuring device data andthe event logbook. No need to reconfigure.
Efficient diagnostics increase measurement reliability• Troubleshooting measures can be called up via the device and in the operating tools• Diverse simulation options, logbook for events that occur and optional line recorderfunctions
Languages

Can be operated in the following languages:• Via local operation
English, German, French, Spanish, Italian, Dutch, Portuguese, Polish, Russian, Turkish,
Chinese, Japanese, Korean, Czech, Swedish• Via web browser
English, German, French, Spanish, Italian, Dutch, Portuguese, Polish, Russian, Turkish,
Chinese, Japanese, Czech, Swedish• Via "FieldCare", "DeviceCare" operating tool: English, German, French, Spanish, Italian,
Chinese, Japanese

Local operation

Via display module
Equipment:• Order code for "Display; operation", option F "4-line, illuminated, graphic display;
touch control""• Order code for "Display; operation", option G "4-line, illuminated, graphic display;
touch control + WLAN"
Information about WLAN interface →  68

A0026785

 37

Operation with touch control

Display elements• 4-line, illuminated, graphic display• White background lighting; switches to red in event of device errors• Format for displaying measured variables and status variables can be individuallyconfigured
Operating elements
External operation via touch control (3 optical keys) without opening the housing: , ,Endress+Hauser

181

Technical data

Proline Teqwave MW 300 HARTVia remote display and operating module DKX001• The remote display and operating module DKX001 is available as an optional extra .
• The measuring device is always supplied with a dummy cover when the remote
display and operating module DKX001 is ordered directly with the measuring
device. Display or operation at the transmitter is not possible in this case.
• If ordered subsequently, the remote display and operating module DKX001 may not
be connected at the same time as the existing measuring device display module.
Only one display or operation unit may be connected to the transmitter at any onetime.

A0026786 38

Operation via remote display and operating module DKX001

Display and operating elements

The display and operating elements correspond to those of the display
module .

Housing material→  180

Cable entry

Corresponds to the selection of the transmitter housing, order code for"Electrical connection"

Connecting cable
Dimensions

on the dimensions: Information"Mechanical construction" section of the "Technical Information"
document.

Remote operation

Via HART protocol
This communication interface is available in device versions with a HART output.

182

Endress+Hauser

Proline Teqwave MW 300 HART

Technical data

8
4

1

2

7

3

5

6
A0028747

 39
1
2
3

4
5
6
7
8

Options for remote operation via HART protocol (active)

Control system (e.g. PLC)
Field Communicator 475
Computer with web browser (e.g. Microsoft Edge) to access the integrated device web server or computer with
operating tool (e.g. FieldCare, DeviceCare, AMS Device Manager, SIMATIC PDM) with COM DTM "CDI
Communication TCP/IP"
Commubox FXA195 (USB)
Field Xpert SFX350 or SFX370
Field Xpert SMT70
VIATOR Bluetooth modem with connecting cable
Transmitter

10
3

2

1

6

4

9

5

7

8
A0028746

 40
1
2
3
4
5

6
7
8
9
10

Endress+Hauser

Options for remote operation via HART protocol (passive)

Control system (e.g. PLC)
Transmitter power supply unit, e.g. RN221N (with communication resistor)
Connection for Commubox FXA195 and Field Communicator 475
Field Communicator 475
Computer with web browser (e.g. Microsoft Edge) to access the integrated device web server or computer with
operating tool (e.g. FieldCare, DeviceCare, AMS Device Manager, SIMATIC PDM) with COM DTM "CDI
Communication TCP/IP"
Commubox FXA195 (USB)
Field Xpert SFX350 or SFX370
Field Xpert SMT70
VIATOR Bluetooth modem with connecting cable
Transmitter

183

Technical data

Service interface

Proline Teqwave MW 300 HART

Via service interface (CDI-RJ45)
A point-to-point connection can be established to configure the device onsite. With the
housing open, the connection is established directly via the service interface (CDI-RJ45) of
the device.
An adapter for the RJ45 to the M12 plug is optionally available:
Order code for "Accessories", option NB: "Adapter RJ45 M12 (service interface)"
The adapter connects the service interface (CDI-RJ45) to an M12 plug mounted in thecable entry. The connection to the service interface can be established via an M12
plug without opening the device.

1
3

2

A0027563

 41
1
2
3

Connection via service interface (CDI-RJ45)

Computer with web browser (e.g. Microsoft Edge) for accessing integrated web server or with "FieldCare"
operating tool, "DeviceCare" with COM DTM "CDI Communication TCP/IP DTM
Standard Ethernet connecting cable with RJ45 plug
Service interface (CDI-RJ45) of the measuring device with access to the integrated Web server

Via WLAN interface
The optional WLAN interface is available on the following device version:
Order code for "Display; operation", option G "4-line, illuminated; touch control + WLAN"
3

1

4

5

6

2

7

3

4

8
A0052607

1
2
3
4
5
6
7
8

184

Transmitter with integrated WLAN antenna
Transmitter with external WLAN antenna
LED lit constantly: WLAN reception is enabled on measuring device
LED flashing: WLAN connection established between operating unit and measuring device
Computer with WLAN interface and web browser (e.g. Microsoft Edge) for accessing the integrated device
web server or with operating tool (e.g. FieldCare, DeviceCare)
Field Xpert SFX350 or SFX370
Field Xpert SMT70
Smartphone or tablet with WLAN interface and web browser (e.g. Microsoft Edge) for accessing the
integrated device web server or with operating tool (e.g. FieldCare, DeviceCare)

Endress+Hauser

Proline Teqwave MW 300 HART

Technical data

Function

WLAN: IEEE 802.11 b/g (2.4 GHz)
• Access point with DHCP server (factory setting)• Network

Encryption

WPA2-PSK AES-128 (in accordance with IEEE 802.11i)

Configurable WLAN channels

1 to 11Degree of protection

IP67

Available antennas• Internal antenna• External antenna (optional)
In the event of poor transmission/reception conditions at the place ofinstallation.
Available as an accessory .

 Only one antenna active in each case!
Range• Internal antenna: typically 10 m (32 ft)• External antenna: typically 50 m (164 ft)

Materials (external antenna)• Antenna: ASA plastic (acrylonitrile-styrene-acrylic ester) and nickelplated brass• Adapter: Stainless steel and nickel-plated brass• Cable: Polyethylene• Plug: Nickel-plated brass• Angle bracket: Stainless steel

Configuring the Internet protocol of the mobile terminal
NOTICE
If the WLAN connection is lost during the configuration, settings made may be lost.
‣ Make sure that the WLAN connection is not disconnected while configuring the device.
NOTICE
Note the following to avoid a network conflict:‣ Avoid accessing the measuring device simultaneously from the same mobile terminal
via the service interface (CDI-RJ45) and the WLAN interface.
‣ Only activate one service interface (CDI-RJ45 or WLAN interface).
‣ If simultaneous communication is necessary: configure different IP address ranges, e.g.
192.168.0.1 (WLAN interface) and 192.168.1.212 (CDI-RJ45 service interface).
Preparing the mobile terminal‣ Enable WLAN on the mobile terminal.
Establishing a WLAN connection from the mobile terminal to the measuring device
1.

In the WLAN settings of the mobile terminal:
Select the measuring device using the SSID (e.g. EH__300_A802000).

2.

If necessary, select the WPA2 encryption method.

3.

Enter the password:
Serial number of the measuring device ex-works (e.g. L100A802000).
 The LED on the display module flashes. It is now possible to operate themeasuring device with the web browser, FieldCare or DeviceCare.
The serial number can be found on the nameplate.
To ensure the safe and swift assignment of the WLAN network to the measuring
point, it is advisable to change the SSID name. It should be possible to clearly assign
the new SSID name to the measuring point (e.g. tag name) because it is displayed as
the WLAN network.

Endress+Hauser

185

Technical data

Proline Teqwave MW 300 HART

Terminating the WLAN connection‣ After configuring the device:
Terminate the WLAN connection between the mobile terminal and measuring device.
Supported operating tools

Different operating tools can be used for local or remote access to the measuring device.
Depending on the operating tool used, access is possible with different operating units andvia a variety of interfaces.
Supported operating
tools

Operating unit

Interface

Additional information

Web browser

Notebook, PC or tablet
with web browser• CDI-RJ45 service
interface• WLAN interface

Special Documentation for thedevice →  192

DeviceCare SFE100

Notebook, PC or tablet• CDI-RJ45 service
with Microsoft Windowsinterfacesystem• WLAN interface• Fieldbus protocol

Service-specific accessories→  164

Notebook, PC or tablet• CDI-RJ45 service
with Microsoft Windowsinterfacesystem• WLAN interface• Fieldbus protocol

Service-specific accessories→  164

FieldCare SFE500

Sources for obtaining device
descriptions
www.endress.com →Download-Area

Sources for obtaining device
descriptions
www.endress.com →Download-Area

Other operating tools based on FDT technology with a device driver such as DTM/
iDTM or DD/EDD can be used for device operation. These operating tools are available
from the individual manufacturers. Integration into the following operating tools,
among others, is supported:• FactoryTalk AssetCentre (FTAC) from Rockwell Automation →www.rockwellautomation.com• Process Device Manager (PDM) from Siemens → www.siemens.com• Asset Management Solutions (AMS) from Emerson → www.emersonprocess.com• FieldCommunicator 375/475 from Emerson → www.emersonprocess.com• Field Device Manager (FDM) from Honeywell → www.process.honeywell.com• FieldMate from Yokogawa → www.yokogawa.com• PACTWare → www.pactware.com
The related device description files are available: www.endress.com → Download Area
Web serverWith the integrated web server, the device can be operated and configured via a web
browser service interface (CDI-RJ45) or WLAN interface. The structure of the operating
menu is the same as for the local display. In addition to the measured values, status
information on the device is displayed and can be used to monitor device health.
Furthermore the device data can be managed and the network parameters can be
configured.
A device that has a WLAN interface (can be ordered as an option) is required for the
WLAN connection: order code for "Display; operation", option G "4-line, illuminated; touch
control + WLAN". The device acts as an Access Point and enables communication bycomputer or a mobile handheld terminal.

186

Endress+Hauser

Proline Teqwave MW 300 HART

Technical data

Supported functions
Data exchange between the operating unit (such as a notebook, for example,) andmeasuring device:• Upload the configuration from the measuring device (XML format, configuration
backup)• Save the configuration to the measuring device (XML format, restore configuration)• Export event list (.csv file)• Export parameter settings (.csv file or PDF file, document the measuring point
configuration)• Export the Heartbeat verification report (PDF file, only available with the Heartbeat
Verification application package)• Flash firmware version for device firmware upgrade, for example• Download driver for system integration• Visualize up to 1000 saved measured values (only available with the Extended
HistoROM application package)
HistoROM
data management

The measuring device features HistoROM data management. HistoROM data management
comprises both the storage and import/export of key device and process data, making
operation and servicing far more reliable, secure and efficient.
When the device is delivered, the factory settings of the configuration data are stored
as a backup in the device memory. This memory can be overwritten with an updated
data record, for example after commissioning.
Additional information on the data storage concept

There are different types of data storage units in which device data are stored and used by the device:
Available data

HistoROM backup

T-DAT

S-DAT

• Event logbook, e.g. diagnostic
events• Parameter data record backup• Device firmware package• Measured value logging ("Extended
HistoROM" order option)• Current parameter data record (used byfirmware at run time)• Indicator (minimum/maximum values)• Totalizer value• Sensor data: e.g. nominal diameter• Serial number• Device configuration (e.g. SW options,
fixed I/O or multi I/O)

Can be plugged into the user interface PC board
in the connection compartment

In the sensor plug in the transmitter neckpart

Storage location Fixed on the user interface PC
board in the connection
compartment

Data backup
Automatic• The most important device data (sensor and transmitter) are automatically saved in the
DAT modules• If the transmitter or measuring device is replaced: once the T-DAT containing theprevious device data has been exchanged, the new measuring device is ready for
operation again immediately without any errors• If exchanging the electronics module (e.g. I/O electronics module): Once the electronics
module has been replaced, the software of the module is compared against the current
device firmware. The module software is upgraded or downgraded where necessary. The
electronics module is available for use immediately afterwards and no compatibility
problems occur.

Endress+Hauser

187

Technical data

Proline Teqwave MW 300 HART

ManualAdditional parameter data record (complete parameter settings) in the integrated device
memory HistoROM backup for:• Data backup function
Backup and subsequent restoration of a device configuration in the device memory
HistoROM backup• Data comparison function
Comparison of the current device configuration with the device configuration saved in
the device memory HistoROM backup
Data transmission
ManualTransfer of a device configuration to another device using the export function of thespecific operating tool, e.g. with FieldCare, DeviceCare or Web server: to duplicate theconfiguration or to store in an archive (e.g. for backup purposes)
Event list
Automatic• Chronological display of up to 20 event messages in the events list
• If the Extended HistoROM application package (order option) is enabled: up to 100
event messages are displayed in the events list along with a time stamp, plain text
description and remedial measures• The events list can be exported and displayed via a variety of interfaces and operating
tools e.g. DeviceCare, FieldCare or Web server
Data logging
ManualIf the Extended HistoROM application package (order option) is enabled:• Recording of 1 to 4 channels of up to 1 000 measured values (up to 250 measured values
per channel)• User configurable recording interval• Export the measured value log via a variety of interfaces and operating tools e.g.
FieldCare, DeviceCare or web server

16.12 Certificates and approvals
Current certificates and approvals for the product are available at www.endress.com on therelevant product page:CE mark

1.

Select the product using the filters and search field.

2.

Open the product page.

3.

Select Downloads.

The device meets the legal requirements of the applicable EU Directives. These are listed in
the corresponding EU Declaration of Conformity along with the standards applied.
Endress+Hauser confirms successful testing of the device by affixing to it the CE mark.

UKCA marking

188

The device meets the legal requirements of the applicable UK regulations (Statutory
Instruments). These are listed in the UKCA Declaration of Conformity along with thedesignated standards. By selecting the order option for UKCA marking, Endress+Hauser
confirms a successful evaluation and testing of the device by affixing the UKCA mark.

Endress+Hauser

Proline Teqwave MW 300 HART

Technical data

Contact address Endress+Hauser UK:
Endress+Hauser Ltd.
Floats Road
Manchester M23 9NF
United Kingdom
www.uk.endress.com
RCM marking

The measuring system meets the EMC requirements of the "Australian Communications
and Media Authority (ACMA)".

Ex-approval

The devices are certified for use in hazardous areas and the relevant safety instructions areprovided in the separate "Safety Instructions" (XA) document. Reference is made to thisdocument on the nameplate.

HART certification

HART interface
The measuring device is certified and registered by the FieldComm Group. The measuringsystem meets all the requirements of the following specifications:• Certified according to HART
• The device can also be operated with certified devices of other manufacturers(interoperability)

Radio approval

The measuring device has radio approval.
For detailed information on the radio approval, see the Special Documentation→  192

Pressure Equipment
Directive

Devices with pressure measuring device approval (Pressure Equipment Directive, PED Cat.
I/II/III) are optionally available: order code for "Additional approval", option LK

Additional certification

Canadian Registration Number (CRN) approval
Devices with Canadian Registration Number (CRN) approval are optionally available: ordercode for "Additional approval", option LD.
Tests and certificates• EN10204-3.1 Material certificate, wetted parts and sensor housing• Pressure test, internal procedure, inspection certificate• EN10204-2.1 Confirmation of compliance with the order and EN10204-2.2 test report

Other standards andguidelines

Endress+Hauser• EN 60529
Degrees of protection provided by enclosures (IP code)• EN 61010-1
Safety requirements for electrical equipment for measurement, control and laboratoryuse - general requirements• EN 61326-1/-2-3
EMC requirements for electrical equipment for measurement, control and laboratory use• ETSI EN 301 489-1/-17
Guidelines for 2.4 GHz radio components• IEC/EN 60068-2-6
Environmental influences: Test procedure - Test Fc: vibration (sinusoidal)• IEC/EN 60068-2-27
Environmental influences: Test procedure - Test Ea: shocks• IEC/EN 60068-2-64
Environmental influences: Test Fh: vibration, broadband random (digital control)
189

Technical data

Proline Teqwave MW 300 HART

• IEC/EN 60068-2-31
Environmental influences: Test procedure - Test Ec: rough handling shocks, primarily for
devices• NAMUR NE 32
Data retention in the event of a power failure in field and control instruments withmicroprocessors• NAMUR NE 43
Standardization of the signal level for the breakdown information of digital transmitterswith analog output signal• NAMUR NE 53
Software of field devices and signal-processing devices with digital electronics• NAMUR NE 105
Specifications for integrating fieldbus devices in engineering tools for field devices• NAMUR NE 107
Self-monitoring and diagnosis of field devices• NAMUR NE 131
Requirements for field devices for standard applications

16.13 Application packages
Many different application packages are available to enhance the functionality of thedevice. Such packages might be needed to address safety aspects or specific application
requirements.
The application packages can be ordered with the device or subsequently fromEndress+Hauser. Detailed information on the order code in question is available from yourlocal Endress+Hauser sales center or on the product page of the Endress+Hauser website:www.endress.com.
Diagnostic functionality

Order code for "Application package", option EA "Extended HistoROM"
Comprises extended functions concerning the event log and the activation of the measured
value memory.
Event log:
Memory volume is extended from 20 message entries (standard version) to up to 100
entries.
Data logging (line recorder):• Memory capacity for up to 1000 measured values is activated.
• 250 measured values can be output via each of the 4 memory channels. The recording
interval can be defined and configured by the user.
• Measured value logs can be accessed via the local display or operating tool e.g. FieldCare,
DeviceCare or Web server.
The application package can also be ordered subsequently: order number DK4009.

Heartbeat Technology

190

Order code for "Application package", option EB "Heartbeat Verification"

Endress+Hauser

Proline Teqwave MW 300 HART

Technical data

Heartbeat Verification
Meets the requirement for traceable verification in accordance with DIN ISO 9001:2008
Clause 7.6 a) "Control of monitoring and measuring equipment"• Functional testing in the installed state without interrupting the process.
• Traceable verification results on request, including a report• Simple testing process via local operation or other operating interfaces.
• Clear measuring point assessment (pass/fail) with high test coverage within theframework of manufacturer specifications.
For detailed information, see the Special Documentation for the device →  192
The application package can also be ordered subsequently: order number DK4009.

16.14 Supplemental documentation
For an overview of the scope of the associated Technical Documentation, refer to thefollowing:• Device Viewer (www.endress.com/deviceviewer): Enter the serial number from thenameplate• Endress+Hauser Operations app: Enter serial number from nameplate or scanmatrix code on nameplate.
Standard documentation

Technical information
DeviceDocumentation code

Proline Teqwave MW 300

TI01763D

Brief Operating Instructions
Brief Operating Instructions for the sensor
SensorDocumentation code

Proline Teqwave MW

KA01671D

Brief Operating Instructions for the transmitter
Transmitter

Documentation code

Proline 300 HART

KA01309D

Description of device parameters

Supplementary devicedependent documentation

Endress+Hauser

DeviceDocumentation code

Proline Teqwave M 300 HART

GP01211D

Safety instructions
Safety instructions for electrical equipment for hazardous areas.
Contents

Documentation code

ATEX: II3G, IECEx: Zone 2

XA03186D

cCSAus: Class I Zone 2, Class I Division 2

XA03188D

191

Technical data

Proline Teqwave MW 300 HART

Special documentation
Contents

Documentation code

Information on the Pressure Equipment Directive

SD01614D

Radio approvals for WLAN interface for A309/A310
display module

SD01793D

Heartbeat Verification application package

SD03168D

Remote display and operating module DKX001

SD01763D

Installation instructions

192

Contents

Note

Installation instructions for spare part sets andaccessories• Call up an overview of all available spare part setsavailable using Device Viewer:
www.endress.com/deviceviewer• Accessories available for order with Installation
Instructions →  162

Endress+Hauser

Proline Teqwave MW 300 HART

Index

Index
A

Access authorization to parameters
Read access . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 58
Write access . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 58
Access code . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 58
Incorrect input . . . . . . . . . . . . . . . . . . . . . . . . . . . 58
Accessories . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 162
Adapting the diagnostic behavior . . . . . . . . . . . . . . . 144
Adapting the status signal . . . . . . . . . . . . . . . . . . . . 144
Additional certification
Canadian Registration Number (CRN) . . . . . . . . . 189
Ambient conditions
Ambient temperature . . . . . . . . . . . . . . . . . . . . . 176
Mechanical load . . . . . . . . . . . . . . . . . . . . . . . . . 177
Operating height . . . . . . . . . . . . . . . . . . . . . . . . 177
Relative humidity . . . . . . . . . . . . . . . . . . . . . . . . 177
Storage temperature . . . . . . . . . . . . . . . . . . . . . . 177
Vibration and shock resistance . . . . . . . . . . . . . . 177
Ambient temperature
Influence . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 176
Ambient temperature range . . . . . . . . . . . . . . . 176, 177
AMS Device Manager . . . . . . . . . . . . . . . . . . . . . . . . 72
Function . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 72
Application . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 166
Applicator . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 164
Approvals . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 188

B

Burst mode . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 77

C

Cable entries
Technical data . . . . . . . . . . . . . . . . . . . . . . . . . . 175
Cable entry
Degree of protection . . . . . . . . . . . . . . . . . . . . . . . 44
Cable specification . . . . . . . . . . . . . . . . . . . . . . . . . . 175
CE mark . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 10, 188
Certificates . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 188
Checklist
Post-connection check . . . . . . . . . . . . . . . . . . . . . 45
Post-mounting check . . . . . . . . . . . . . . . . . . . . . . 32
Cleaning
Exterior cleaning . . . . . . . . . . . . . . . . . . . . . . . . 159
Commissioning . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 80
Adjusting the measured value based on thereference value . . . . . . . . . . . . . . . . . . . . . . . . . . . 99
Advanced settings . . . . . . . . . . . . . . . . . . . . . . . 100
Basic settings for the adjustment . . . . . . . . . . . . . 99
Configuring the measuring device . . . . . . . . . . . . . 81
Communication-specific data . . . . . . . . . . . . . . . . . . . 75
Connecting cable . . . . . . . . . . . . . . . . . . . . . . . . . . . . 34
Connecting the measuring device . . . . . . . . . . . . . . . . 35
Connecting the signal cables . . . . . . . . . . . . . . . . . . . 36
Connecting the supply voltage cables . . . . . . . . . . . . . 36
Connection
see Electrical connection
Endress+Hauser

Connection cable . . . . . . . . . . . . . . . . . . . . . . . . . . . . 33
Connection preparations . . . . . . . . . . . . . . . . . . . . . . 35
Connection tools . . . . . . . . . . . . . . . . . . . . . . . . . . . . 33
Context menu
Calling up . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 55
Closing . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 55
Explanation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 55
Current consumption . . . . . . . . . . . . . . . . . . . . . . . . 174

D

Date of manufacture . . . . . . . . . . . . . . . . . . . . . . 16, 17
Declaration of Conformity . . . . . . . . . . . . . . . . . . . . . 10
Defining the access code . . . . . . . . . . . . . . . . . . . . . 116
Degree of protection . . . . . . . . . . . . . . . . . . . . . 44, 177
DesignMeasuring device . . . . . . . . . . . . . . . . . . . . . . . . . 13
Operating menu . . . . . . . . . . . . . . . . . . . . . . . . . . 47
Device components . . . . . . . . . . . . . . . . . . . . . . . . . . 13
Device description files . . . . . . . . . . . . . . . . . . . . 73, 74
Device locking, status . . . . . . . . . . . . . . . . . . . . . . . 119
Device nameSensor . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 17
Transmitter . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 16
Device repair . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 160
Device revision . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 73
Device type code . . . . . . . . . . . . . . . . . . . . . . . . . . . . 73
Device Viewer . . . . . . . . . . . . . . . . . . . . . . . . . . 15, 160
Device-specific accessories . . . . . . . . . . . . . . . . . . . . 162
DeviceCare . . . . . . . . . . . . . . . . . . . . . . . . . . . . 71, 164
Device description file . . . . . . . . . . . . . . . . . . . . . . 74
Diagnosis
Symbols . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 137
Diagnostic behavior
Explanation . . . . . . . . . . . . . . . . . . . . . . . . . . . . 138
Symbols . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 138
Diagnostic information
Design, description . . . . . . . . . . . . . . . . . . . 138, 142
DeviceCare . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 142
FieldCare . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 142
LEDs . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 136
Local display . . . . . . . . . . . . . . . . . . . . . . . . . . . 137
Overview . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 146
Remedial measures . . . . . . . . . . . . . . . . . . . . . . 146
Web browser . . . . . . . . . . . . . . . . . . . . . . . . . . . 140
Diagnostic message . . . . . . . . . . . . . . . . . . . . . . . . . 137
Diagnostics list . . . . . . . . . . . . . . . . . . . . . . . . . . . . 150
Dimensions . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 24
DIP switch
see Write protection switch
Disabling write protection . . . . . . . . . . . . . . . . . . . . 116
Displaysee Local display
Display and operating module DKX001 . . . . . . . . . . 182
Display area
For operational display . . . . . . . . . . . . . . . . . . . . . 49
In the navigation view . . . . . . . . . . . . . . . . . . . . . 51
193

Index

Display values
For locking status . . . . . . . . . . . . . . . . . . . . . . . . 119
Displaying the measured value history . . . . . . . . . . . 127
Disposal . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 161
Document
Function . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6
Symbols . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6
Document function . . . . . . . . . . . . . . . . . . . . . . . . . . . 6

E

Editing view . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 53
Input screen . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 54
Using operating elements . . . . . . . . . . . . . . . . 53, 54
Electrical conductivity . . . . . . . . . . . . . . . . . . . 167, 178
Electrical connection
Commubox FXA195 (USB) . . . . . . . . . . . . . . 66, 182
Computer with web browser (e.g. Microsoft Edge). . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 66, 182
Degree of protection . . . . . . . . . . . . . . . . . . . . . . . 44
Field Communicator 475 . . . . . . . . . . . . . . . 66, 182
Field Xpert SFX350/SFX370 . . . . . . . . . . . . . 66, 182
Field Xpert SMT70 . . . . . . . . . . . . . . . . . . . . 66, 182
Measuring device . . . . . . . . . . . . . . . . . . . . . 33, 174
Operating tool (e.g. FieldCare, AMS Device
Manager, SIMATIC PDM) . . . . . . . . . . . . . . . 66, 182
Operating tools
Via HART protocol . . . . . . . . . . . . . . . . . . 66, 182
Via service interface (CDI-RJ45) . . . . . . . . 68, 184
Via WLAN interface . . . . . . . . . . . . . . . . 68, 184
VIATOR Bluetooth modem . . . . . . . . . . . . . . 66, 182
Web server . . . . . . . . . . . . . . . . . . . . . . . . . . 68, 184
WLAN interface . . . . . . . . . . . . . . . . . . . . . . 68, 184
Electromagnetic compatibility . . . . . . . . . . . . . . . . . 177
Electronics module . . . . . . . . . . . . . . . . . . . . . . . . . . 13
Enabling write protection . . . . . . . . . . . . . . . . . . . . 116
Enabling/disabling the keypad lock . . . . . . . . . . . . . . 59
Endress+Hauser services
Maintenance . . . . . . . . . . . . . . . . . . . . . . . . . . . 159
Repair . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 160
Error messages
see Diagnostic messages
Event logbook . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 152
Events list . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 152
Ex connection data . . . . . . . . . . . . . . . . . . . . . . . . . 173
Ex-approval . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 189
Extended order code
Sensor . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 17
Transmitter . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 16Exterior cleaning . . . . . . . . . . . . . . . . . . . . . . . . . . . 159

F

Field Communicator
Function . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 72
Field Communicator 475 . . . . . . . . . . . . . . . . . . . . . . 72
Field of application
Residual risks . . . . . . . . . . . . . . . . . . . . . . . . . . . . 10
Field Xpert
Function . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 70
Field Xpert SFX350 . . . . . . . . . . . . . . . . . . . . . . . . . . 70
194

Proline Teqwave MW 300 HART

FieldCare . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 70, 164
Device description file . . . . . . . . . . . . . . . . . . . . . . 74
Establishing a connection . . . . . . . . . . . . . . . . . . . 71
Function . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 70
User interface . . . . . . . . . . . . . . . . . . . . . . . . . . . 71
Filtering the event logbook . . . . . . . . . . . . . . . . . . . 152
FirmwareRelease date . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 73
Version . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 73
Firmware history . . . . . . . . . . . . . . . . . . . . . . . . . . . 158
Flow direction . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 23
Flow limit . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 178
Function range
AMS Device Manager . . . . . . . . . . . . . . . . . . . . . . 72
SIMATIC PDM . . . . . . . . . . . . . . . . . . . . . . . . . . . 72
Function scope
Field Communicator . . . . . . . . . . . . . . . . . . . . . . 72
Field Communicator 475 . . . . . . . . . . . . . . . . . . . 72
Field Xpert . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 70
Functionssee Parameters

G

Galvanic isolation . . . . . . . . . . . . . . . . . . . . . . . . . . 173

H

Hardware write protection . . . . . . . . . . . . . . . . . . . . 117
HART certification . . . . . . . . . . . . . . . . . . . . . . . . . . 189
HART protocol
Device variables . . . . . . . . . . . . . . . . . . . . . . . . . . 75
Dynamic variables . . . . . . . . . . . . . . . . . . . . . . . . 75
Help text
Calling up . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 57
Closing . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 57
Explanation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 57
HistoROM . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 109

I

Identifying the measuring device . . . . . . . . . . . . . . . . 15
Incoming acceptance . . . . . . . . . . . . . . . . . . . . . . . . . 14
Indication
Current diagnostic event . . . . . . . . . . . . . . . . . . . 150
Previous diagnostic event . . . . . . . . . . . . . . . . . . 150
Influence
Ambient temperature . . . . . . . . . . . . . . . . . . . . . 176
Information about this document . . . . . . . . . . . . . . . . . 6
Inlet runs . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 24
Input . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 166
Inspection
Connection . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 45
Mounting procedure . . . . . . . . . . . . . . . . . . . . . . . 32
Received goods . . . . . . . . . . . . . . . . . . . . . . . . . . . 14
Installation point . . . . . . . . . . . . . . . . . . . . . . . . . . . . 21
Intended use . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9

LLanguages, operation options . . . . . . . . . . . . . . . . . 181
Line recorder . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 127
Local display . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 181
Navigation view . . . . . . . . . . . . . . . . . . . . . . . . . . 51
Endress+Hauser

Proline Teqwave MW 300 HART

see Diagnostic message
see In alarm condition
see Operational display
Text editor . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 53

M

Main electronics module . . . . . . . . . . . . . . . . . . . . . . 13
Maintenance work . . . . . . . . . . . . . . . . . . . . . . . . . 159
Managing the device configuration . . . . . . . . . . . . . 109
Manufacturer ID . . . . . . . . . . . . . . . . . . . . . . . . . . . . 73
Materials . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 179
Measured variable . . . . . . . . . . . . . . . . . . . . . . . . . . 166
Measuring device
Adjusting the measured value based on thereference value . . . . . . . . . . . . . . . . . . . . . . . . . . . 99
Basic settings for the adjustment . . . . . . . . . . . . . 99
Configuring . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 81
Conversion . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 160
Design . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 13
Disposal . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 161
Integrating via communication protocol . . . . . . . . 73
Mounting the sensor . . . . . . . . . . . . . . . . . . . . . . 27
Screw tightening torques . . . . . . . . . . . . . . . . . 28
Preparing for electrical connection . . . . . . . . . . . . 35
Preparing for mounting . . . . . . . . . . . . . . . . . . . . 27
Removing . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 161
Repairs . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 160
Switching on . . . . . . . . . . . . . . . . . . . . . . . . . . . . 80
Measuring principle . . . . . . . . . . . . . . . . . . . . . . . . 166
Measuring range . . . . . . . . . . . . . . . . . . . . . . . . . . . 167
Measuring system . . . . . . . . . . . . . . . . . . . . . . . . . . 166
Mechanical load . . . . . . . . . . . . . . . . . . . . . . . . . . . 177
Medium temperature range . . . . . . . . . . . . . . . . . . . 178
Menu
Diagnostics . . . . . . . . . . . . . . . . . . . . . . . . . . . . 150
Setup . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 82
Menus
For measuring device configuration . . . . . . . . . . . . 81
For specific settings . . . . . . . . . . . . . . . . . . . . . . 100
Mounting dimensions
see Dimensions
Mounting kit . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 163
Mounting preparations . . . . . . . . . . . . . . . . . . . . . . . 27
Mounting procedure . . . . . . . . . . . . . . . . . . . . . . . . . 21
Mounting requirements
Dimensions . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 24
Inlet and outlet runs . . . . . . . . . . . . . . . . . . . . . . . 24
Installation point . . . . . . . . . . . . . . . . . . . . . . . . . 21
Orientation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 23
Static pressure . . . . . . . . . . . . . . . . . . . . . . . . . . 178
Thermal insulation . . . . . . . . . . . . . . . . . . . . . . . . 25
Vibrations . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 178

N

NameplateSensor . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 17
Transmitter . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 16
Navigation path (navigation view) . . . . . . . . . . . . . . . 51

Endress+Hauser

Index

Navigation view
In the submenu . . . . . . . . . . . . . . . . . . . . . . . . . . 51
In the wizard . . . . . . . . . . . . . . . . . . . . . . . . . . . . 51
Netilion . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 164
Numeric editor . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 53

O

Onsite display
Numeric editor . . . . . . . . . . . . . . . . . . . . . . . . . . . 53
Operating elements . . . . . . . . . . . . . . . . . . . . . . 55, 138
Operating height . . . . . . . . . . . . . . . . . . . . . . . . . . . 177
Operating keys
see Operating elements
Operating menu
Design . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 47
Menus, submenus . . . . . . . . . . . . . . . . . . . . . . . . 47
Submenus and user roles . . . . . . . . . . . . . . . . . . . 48
Operating philosophy . . . . . . . . . . . . . . . . . . . . . . . . 48
Operation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 119
Operation options . . . . . . . . . . . . . . . . . . . . . . . . . . . 46
Operational display . . . . . . . . . . . . . . . . . . . . . . . . . . 49
Operational safety . . . . . . . . . . . . . . . . . . . . . . . . . . . 10
Order code . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 16, 17
Orientation (vertical, horizontal) . . . . . . . . . . . . . . . . 23
Outlet runs . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 24
Output signal . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 169
Output variables . . . . . . . . . . . . . . . . . . . . . . . . . . . 169

P

Packaging disposal . . . . . . . . . . . . . . . . . . . . . . . . . . 20
Parameter
Changing . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 58
Entering values or text . . . . . . . . . . . . . . . . . . . . . 58
Parameter settings
Administration (Submenu) . . . . . . . . . . . . . . . . . 112
Advanced setup (Submenu) . . . . . . . . . . . . . . . . 101
Burst configuration 1 to n (Submenu) . . . . . . . . . . 77
Configuration backup (Submenu) . . . . . . . . . . . . 109
Current input . . . . . . . . . . . . . . . . . . . . . . . . . . . . 85
Current input (Wizard) . . . . . . . . . . . . . . . . . . . . . 85
Current input 1 to n (Submenu) . . . . . . . . . . . . . 121
Current output . . . . . . . . . . . . . . . . . . . . . . . . . . . 87
Current output (Wizard) . . . . . . . . . . . . . . . . . . . . 87
Data logging (Submenu) . . . . . . . . . . . . . . . . . . . 127
Define access code (Wizard) . . . . . . . . . . . . . . . . 111
Device information (Submenu) . . . . . . . . . . . . . . 156
Diagnostics (Menu) . . . . . . . . . . . . . . . . . . . . . . 150
Display (Submenu) . . . . . . . . . . . . . . . . . . . . . . . 103
Display (Wizard) . . . . . . . . . . . . . . . . . . . . . . . . . 96
I/O configuration . . . . . . . . . . . . . . . . . . . . . . . . . 84
I/O configuration (Submenu) . . . . . . . . . . . . . . . . 84
Output (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 75
Process variables (Submenu) . . . . . . . . . . . . . . . . 120
Pulse/frequency/switch output . . . . . . . . . . . . . . . 89
Pulse/frequency/switch output (Wizard) . . 89, 90, 93
Pulse/frequency/switch output 1 to n (Submenu) 123
Relay output . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 95
Relay output 1 to n (Submenu) . . . . . . . . . . . . . . 123
Relay output 1 to n (Wizard) . . . . . . . . . . . . . . . . . 95
195

IndexReset access code (Submenu) . . . . . . . . . . . . . . . 111
Setup (Menu) . . . . . . . . . . . . . . . . . . . . . . . . . . . . 82
Simulation (Submenu) . . . . . . . . . . . . . . . . . . . . 113
Status input . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 86
Status input 1 to n (Submenu) . . . . . . . . . . . . . . 121
Status input 1 to n (Wizard) . . . . . . . . . . . . . . . . . 86
System units (Submenu) . . . . . . . . . . . . . . . . . . . . 83
Totalizer (Submenu) . . . . . . . . . . . . . . . . . . . . . . 124
Totalizer 1 to n (Submenu) . . . . . . . . . . . . . . . . . 102
Totalizer handling (Submenu) . . . . . . . . . . . . . . . 125
Value current output 1 to n (Submenu) . . . . . . . . 122
Web server (Submenu) . . . . . . . . . . . . . . . . . . . . . 65
WLAN settings (Wizard) . . . . . . . . . . . . . . . . . . 107
Performance characteristics . . . . . . . . . . . . . . . . . . . 175
Post-connection check . . . . . . . . . . . . . . . . . . . . . . . . 80
Post-connection check (checklist) . . . . . . . . . . . . . . . 45
Post-mounting check . . . . . . . . . . . . . . . . . . . . . . . . . 80
Post-mounting check (checklist) . . . . . . . . . . . . . . . . 32
Potential equalization . . . . . . . . . . . . . . . . . . . . 39, 174
Power consumption . . . . . . . . . . . . . . . . . . . . . . . . . 174
Power supply failure . . . . . . . . . . . . . . . . . . . . . . . . 174
Pressure Equipment Directive . . . . . . . . . . . . . . . . . 189
Pressure Equipment Directive (PED) . . . . . . . . . . . . . 189
Pressure/temperature ratings . . . . . . . . . . . . . . . . . 178
Process conditions
Flow limit . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 178
Medium temperature . . . . . . . . . . . . . . . . . . . . . 178
Process variable . . . . . . . . . . . . . . . . . . . . . . . . . . . . 166
Product safety . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 10
Protecting parameter settings . . . . . . . . . . . . . . . . . 116

RRadio approval . . . . . . . . . . . . . . . . . . . . . . . . . . . . 189
RCM marking . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 189
Read access . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 58
Reading off measured values . . . . . . . . . . . . . . . . . . 120
Registered trademarks . . . . . . . . . . . . . . . . . . . . . . . . . 8
Remedial measures
Calling up . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 139
Closing . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 139
Remote operation . . . . . . . . . . . . . . . . . . . . . . . . . . 182
Repair . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 160
Notes . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 160
Repair of a device . . . . . . . . . . . . . . . . . . . . . . . . . . 160
Repeatability . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 176
Replacement
Device components . . . . . . . . . . . . . . . . . . . . . . . 160
Requirements for personnel . . . . . . . . . . . . . . . . . . . . . 9
Retrofit kit for display/WLAN . . . . . . . . . . . . . . . . . 164
Retrofit kit for inputs/outputs . . . . . . . . . . . . . . . . . 164
Return . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 160

S

Safety . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9Screw tightening torques . . . . . . . . . . . . . . . . . . . . . . 28
SensorMounting . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 27
Serial number . . . . . . . . . . . . . . . . . . . . . . . . . . . 16, 17
Setting the operating language . . . . . . . . . . . . . . . . . 80
196

Proline Teqwave MW 300 HART

Settings
Adapting the measuring device to the process
conditions . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 119
Administration . . . . . . . . . . . . . . . . . . . . . . . . . . 111
Advanced display configurations . . . . . . . . . . . . . 103
Current input . . . . . . . . . . . . . . . . . . . . . . . . . . . . 85
Current output . . . . . . . . . . . . . . . . . . . . . . . . . . . 87
I/O configuration . . . . . . . . . . . . . . . . . . . . . . . . . 84
Local display . . . . . . . . . . . . . . . . . . . . . . . . . . . . 96
Managing the device configuration . . . . . . . . . . . 109
Operating language . . . . . . . . . . . . . . . . . . . . . . . 80
Pulse output . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 89
Pulse/frequency/switch output . . . . . . . . . . . . 89, 90
Relay output . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 95
Resetting the device . . . . . . . . . . . . . . . . . . . . . . 155
Resetting the totalizer . . . . . . . . . . . . . . . . . . . . 125
Simulation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 113
Status input . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 86
Switch output . . . . . . . . . . . . . . . . . . . . . . . . . . . . 93
System units . . . . . . . . . . . . . . . . . . . . . . . . . . . . 83
Tag name . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 82
Totalizer . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 102
Totalizer reset . . . . . . . . . . . . . . . . . . . . . . . . . . 125
WLAN . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 107
Signal on alarm . . . . . . . . . . . . . . . . . . . . . . . . . . . . 171
SIMATIC PDM . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 72
Function . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 72
Software release . . . . . . . . . . . . . . . . . . . . . . . . . . . . 73
Spare part . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 160
Spare parts . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 160
Special connection instructions . . . . . . . . . . . . . . . . . 40
Standards and guidelines . . . . . . . . . . . . . . . . . . . . . 189
Static pressure . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 178
Status area
For operational display . . . . . . . . . . . . . . . . . . . . . 49
In the navigation view . . . . . . . . . . . . . . . . . . . . . 51
Status signals . . . . . . . . . . . . . . . . . . . . . . . . . 137, 140
Storage concept . . . . . . . . . . . . . . . . . . . . . . . . . . . . 187
Storage conditions . . . . . . . . . . . . . . . . . . . . . . . . . . . 19
Storage temperature . . . . . . . . . . . . . . . . . . . . . . . . . 19
Storage temperature range . . . . . . . . . . . . . . . . . . . 177
Submenu
Administration . . . . . . . . . . . . . . . . . . . . . . 111, 112
Advanced setup . . . . . . . . . . . . . . . . . . . . . 100, 101
Burst configuration 1 to n . . . . . . . . . . . . . . . . . . . 77
Configuration backup . . . . . . . . . . . . . . . . . . . . . 109
Current input 1 to n . . . . . . . . . . . . . . . . . . . . . . 121
Data logging . . . . . . . . . . . . . . . . . . . . . . . . . . . 127
Device information . . . . . . . . . . . . . . . . . . . . . . . 156
Display . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 103
Events list . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 152
Heartbeat setup . . . . . . . . . . . . . . . . . . . . . . . . . 108
I/O configuration . . . . . . . . . . . . . . . . . . . . . . . . . 84
Input values . . . . . . . . . . . . . . . . . . . . . . . . . . . . 121
Measured values . . . . . . . . . . . . . . . . . . . . . . . . 120
Output . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 75
Output values . . . . . . . . . . . . . . . . . . . . . . . . . . . 122
Overview . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 48
Process variables . . . . . . . . . . . . . . . . . . . . . . . . 120
Endress+Hauser

Proline Teqwave MW 300 HART

Pulse/frequency/switch output 1 to n . . . . . . . . . 123
Relay output 1 to n . . . . . . . . . . . . . . . . . . . . . . . 123
Reset access code . . . . . . . . . . . . . . . . . . . . . . . . 111
Simulation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 113
Status input 1 to n . . . . . . . . . . . . . . . . . . . . . . . 121
System units . . . . . . . . . . . . . . . . . . . . . . . . . . . . 83
Total solids adjustment . . . . . . . . . . . . . . . . . . . . . 99
Totalizer . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 124
Totalizer 1 to n . . . . . . . . . . . . . . . . . . . . . . . . . . 102
Totalizer handling . . . . . . . . . . . . . . . . . . . . . . . 125
Value current output 1 to n . . . . . . . . . . . . . . . . . 122
Web server . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 65
Supplemental documentation . . . . . . . . . . . . . . . . . 191
Supply voltage . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 174
Switch output . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 171
Symbols
Controlling data entries . . . . . . . . . . . . . . . . . . . . 54
For communication . . . . . . . . . . . . . . . . . . . . . . . 49
For diagnostic behavior . . . . . . . . . . . . . . . . . . . . 49
For locking . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 49
For measured variable . . . . . . . . . . . . . . . . . . . . . 49
For measurement channel number . . . . . . . . . . . . 49
For menus . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 51
For parameters . . . . . . . . . . . . . . . . . . . . . . . . . . . 51
For status signal . . . . . . . . . . . . . . . . . . . . . . . . . . 49
For submenu . . . . . . . . . . . . . . . . . . . . . . . . . . . . 51
For wizards . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 51
In the status area of the local display . . . . . . . . . . . 49
Input screen . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 54
Operating elements . . . . . . . . . . . . . . . . . . . . . . . 53
System design
Measuring system . . . . . . . . . . . . . . . . . . . . . . . 166
see Measuring device design
System integration . . . . . . . . . . . . . . . . . . . . . . . . . . 73

T

Technical data, overview . . . . . . . . . . . . . . . . . . . . . 166
Temperature range
Ambient temperature for display . . . . . . . . . . . . 181
Storage temperature . . . . . . . . . . . . . . . . . . . . . . . 19
Terminal assignment . . . . . . . . . . . . . . . . . . . . . . . . . 35
Terminals . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 174
Text editor . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 53
Thermal insulation . . . . . . . . . . . . . . . . . . . . . . . . . . 25
Tool tipsee Help text
Tools
Electrical connection . . . . . . . . . . . . . . . . . . . . . . 33
For transport . . . . . . . . . . . . . . . . . . . . . . . . . . . . 19
Totalizer
Configuring . . . . . . . . . . . . . . . . . . . . . . . . . . . . 102
Transmitter
Turning the display module . . . . . . . . . . . . . . . . . 30
Turning the housing . . . . . . . . . . . . . . . . . . . . . . . 29
Transporting the measuring device . . . . . . . . . . . . . . 19
Troubleshooting
General . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 134
Turning the display module . . . . . . . . . . . . . . . . . . . . 30

Endress+Hauser

Index

Turning the electronics housing
see Turning the transmitter housing
Turning the transmitter housing . . . . . . . . . . . . . . . . 29

U

UKCA marking . . . . . . . . . . . . . . . . . . . . . . . . . . . . 188
Use of measuring device
Borderline cases . . . . . . . . . . . . . . . . . . . . . . . . . . . 9
Incorrect use . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9
see Intended use
User roles . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 48

V

Version data for the device . . . . . . . . . . . . . . . . . . . . 73
Vibration and shock resistance . . . . . . . . . . . . . . . . . 177
Vibrations . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 178

W

WeightCompact version
SI units . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 178
Transport (notes) . . . . . . . . . . . . . . . . . . . . . . . . . 19
WizardCurrent input . . . . . . . . . . . . . . . . . . . . . . . . . . . . 85
Current output . . . . . . . . . . . . . . . . . . . . . . . . . . 87
Define access code . . . . . . . . . . . . . . . . . . . . . . . 111
Display . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 96
Pulse/frequency/switch output . . . . . . . . . 89, 90, 93
Relay output 1 to n . . . . . . . . . . . . . . . . . . . . . . . . 95
Status input 1 to n . . . . . . . . . . . . . . . . . . . . . . . . 86
Total solids commissioning . . . . . . . . . . . . . . . . . . 99
WLAN settings . . . . . . . . . . . . . . . . . . . . . . . . . . 107
WLAN settings . . . . . . . . . . . . . . . . . . . . . . . . . . . . 107
Workplace safety . . . . . . . . . . . . . . . . . . . . . . . . . . . 10
Write access . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 58
Write protection
Via access code . . . . . . . . . . . . . . . . . . . . . . . . . . 116
Via write protection switch . . . . . . . . . . . . . . . . . 117
Write protection switch . . . . . . . . . . . . . . . . . . . . . . 117

197

*71637042*
71637042

www.addresses.endress.com