No. SS2-MGG200-0100

MagneW™ PLUS+
Electromagnetic Flowmeter Detector(General, FM Nonincendive Approval)
Model MGG18/MGG19/MGG11OVERVIEW
The MagneW PLUS+ electromagnetic flowmeter detector
is a high performance, highly reliable flowmeter developedwith Azbil Corporation’s proven MagneW3000 flow measurement technologies. Model MGG18 (watertight model)
and model MGG19 (submersible model) offer superior
process flowrate measurement and couple with a wide rangeof MagneW PLUS+ converters.

Size 2.5 to 600 mm (0.1 to 24 inches)

FEATURESHigh performance lining• A new, exclusive high quality lining technology and a special mirror-finish PFA lining offers higher anti-adhesive
properties than existing models.

• The mirror-finish PFA lining is particularly applicable forSize 700 to 1100 mm (28 to 44 inches)measurement of sticky pulp and gypsum slurries.

• Only pure white PFA with no additives is used to makeA wide variety of piping connections• A hose or union joint or clamp can be selected for very

new linings.

• The successful embedded punch plate that offers provensmall size models (diameters of 2.5 to 15 mm (0.1 to 1/2
inches)).

performance under conditions such as rapid thermalchange and negative pressure. PFA linings with diameter
ranges from 2.5 mm to 600 mm (0.1 to 24 inches) areavailable, making selection of the best lining easy for awide variety of applications.

• A flange structure is available for all sizes (sizes of 2.5 to1100 mm (0.1 to 44 inches).

• A wafer construction can be also selected (sizes of 2.5 to
200 mm (0.1 to 8 inches)).

Replacement interfacing detector (optional)• This detector can replace the detector interfaces of our• Sizes of 65 and 125 mm (2½ and 5 inches) have been
added to our existing product lineup.

existing models and those of other manufacturers. Please
consult an Azbil Corp. representative for details.

Compatibility• Remote model converters can be used in combination

Rugged detector structure• A stainless steel case has been adopted for sizes of 2.5 mm

with our conventional converters. Please consult an Azbil
Corp. representative for details.

to 200 mm (0.1 to 8 inches).

• A watertight structure effective for environments where

Type of protection

moisture and condensation tends to occur is used for thewater-tight model (model MGG18).

Model MGG 18/19 are suitable for use in FM Nonincendive
Class I, Division 2, Groups A, B, C and D; Class II, Division 2,
Groups F and G; Class III, Division 2.

Improved Accuracy Specification
The standard accuracy is +/- 0.5 % of rate. Also available
is an optional high accuracy calibration rated at +/- 0.35 %of rate (sizes of 40 mm to 350 mm (1½ to 14 inches), combined with MGG14C).
Specifications are subject to change without notice.

1“This product is designed for general industrial use.”

18th edition

No. SS2-MGG200-0100

Azbil Corporation

APPLICATIONS

FM approval for MGG18 and MGG19
Nonincendive for Class I, Division 2, Groups A, B, C and D
Suitable for Class II, Division 2, Groups F and G
Suitable for Class III, Division 2, indoor and outdoor (type4X) hazardous locations.

Pulp and paper
Pulp liquids, chemicals, corrosive liquids, industrial water,
wastewater, etc.

Petroleum/petrochemical/chemicals

European Pressure Equipment Directive(2014/68/EU)

Corrosive liquids, dyestuffs, chemicals, industrial water,
waste water, etc.

This product is subject to the European Pressure Equipment
Directive (PED).

Public utilities

Article 4 of the PED differentiates pressure equipment according to the degree of danger.

Water supply systems, sewage systems, community drainage,
human waste, sludge, sediment slurry, regulation of total
effluent, etc.

The maximum allowable pressure of this product is statedon page 5 of this document. Note, however, that because
this product is designed and manufactured in accordancewith sound engineering practice (SEP) as described inarticle 4, section 3 of the PED, there are restrictions on thepressure range when this product is used in a country where
PED is applicable.

Food
Potable water, light, medium and high density fluids, industrial water, waste water, etc.

Steel/nonferrous metals/ceramics
Aluminum slurry, cooling water, industrial water, corrosive
liquids, wastewater, etc.

Determine the maximum allowable pressure by checking thefollowing items.

Machinery/equipment/electric machinery(1) Group of the fluid

Corrosive liquids, cooking water, circulating water, industrial water, wastewater, etc.

Check the group of the fluid according to article 13 of
the PED.

Construction• Group 1: Hazardous fluids

Building material slurry, sediment slurry, cement slurry,industrial water, etc.

• Group 2: Non-hazardous fluids(2) Vapor pressure at the maximum allowable temperatureof the measured fluid

Shipbuilding
Sediment slurry etc.

Check the applicable category, (i) or (ii).
(i) Liquid whose vapor pressure at the maximum allowable temperature is greater than 0.5 bar above normal
atmospheric pressure (1013 mbar)

Electric power
Corrosive liquids, cooling water, industrial water, wastewater, etc.

(ii) Liquid having a vapor pressure at the maximum allowable temperature of not more than 0.5 bar above
normal atmospheric pressure (1013 mbar)

Gas
Circulating water for air conditioning, etc.

(3) Nominal size (DN) of the electromagnetic flowmeter

FUNCTIONAL SPECIFICATIONSCheck the nominal size of the flowmeter.

Type of protection(4) Maximum allowable pressure for equipment designed by
SEP.

Model MGG18, MGG11

In table 1, find the cell where the results of (1), (2), and(3) meet.

JIS C 0920 watertight model
NEMA ICS6-110 TYPE4X
IEC PUBL 529 IP67

“Tables 6–9” shown in table 1 below are taken fromarticle 4 and annex II of the PED.

Model MGG19(5) Maximum pressure

JIS C 0920 submersible model
NEMA ICS6-110 TYPE6
IEC PUBL 529 IP68

Whichever of the pressures below is the lowest is the applicable pressure.
• Maximum pressure for this product: see page 5 of

Note: The performance of the submersible model was evaluated bysinking it 1 m below the surface of contaminated water for 1
month.
If the product will be submerged for a long consecutive periodof time or in a corrosive fluid, please contact us.

this document• Maximum pressure for SEP equipment defined by the
PED: see (4) above• Maximum pressure for the flange: see the applicablestandard
2

Azbil Corporation

No. SS2-MGG200-0100
Table 1: Maximum allowable pressure for SEP products(1) Fluid group(2) Vapor pressure
PED table
mm(3)
Nominal size
(DN)

2.5
5
10
15
25
40
50
65
80
100
125
150
200
250
300
350
400
450
500
600
700
800
900
1000
1100

Group 1(i)
Table 6bar
No limit
No limit
No limit
No limit
No limit
0.5
0.5
0.5
0.5
0.5
0.5
0.5
0.5
0.5
0.5
0.5
0.5
0.5
0.5
0.5
0.5
0.5
0.5
0.5
0.5

MPa
No limit
No limit
No limit
No limit
No limit
0.05
0.05
0.05
0.05
0.05
0.05
0.05
0.05
0.05
0.05
0.05
0.05
0.05
0.05
0.05
0.05
0.05
0.05
0.05
0.05

Group 2
Group 1(i)(ii)
Table 7
Table 8(4) Maximum allowable pressure
bar
MPa
bar
MPa
No limit
No limit
No limit
No limit
No limit
No limit
No limit
No limit
No limit
No limit
No limit
No limit
No limit
No limit
No limit
No limit
No limit
No limit
No limit
No limit
25.0
2.50 No limit
No limit
20.0
2.00 No limit
No limit
15.3
1.53 No limit
No limit
12.5
1.25
25.0
2.50
10.0
1.00
20.0
2.00
8.0
0.80
16.0
1.60
6.6
0.66
13.3
1.33
5.0
0.50
10.0
1.00
4.0
0.40
8.0
0.80
3.3
0.33
6.6
0.66
2.8
0.28
5.7
0.57
2.5
0.25
5.0
0.50
2.2
0.22
4.4
0.44
2.0
0.20
4.0
0.40
1.6
0.16
3.3
0.33
1.4
0.14
2.8
0.28
1.2
0.12
2.5
0.25
1.1
0.11
2.2
0.22
1.0
0.10
2.0
0.20
0.9
0.09
1.8
0.18

Line size
2.5, 5, 10, 15, 25, 40, 50, 65, 80, 100, 125, 150, 200, 250, 300,
350, 400, 450, 500, 600, 700, 800, 900, 1000, 1100 mm(0.1, 0.2, 3/8, 1/2, 1, 1½, 2, 2½, 3, 4, 5, 6, 8, 10, 12, 14, 16, 18,
20, 24, 28, 32, 36, 40, 44 inches)

Flange rating
JIS 10K, JIS 16K, JIS 20K, JIS 30K,
JPI 150, JPI 300, ANSI 150, ANSI 300,
DIN PN10, DIN PN16, DIN PN25, DIN PN40
(Size 2.5 to 65 mm (0.1 to 2.5 inches))
JIS 10K, JIS 16K, JIS 20K, JIS 30K, JIS G3443-2 F12
JPI 150, JPI 300,ANSI 150, ANSI 300,
DIN PN10, DIN PN16, DIN PN25, DIN PN40
(Size 80 to 200 mm (3 to 8 inches))
JIS 10K, JIS 16K, JIS 20K, JIS G3443-2 F12
JPI 150, JPI 300, ANSI 150, ANSI 300,
DIN PN10, DIN PN16, DIN PN25
(Size 250 to 600 mm (10 to 24 inches), PFA/ETFE lining)
JIS 10K, JIS G3443-2 F12,
JPI 150, ANSI 150, DIN PN10(Size 700 to 1100 mm (28 to 44 inches), chloroprene rubber
lining)Reference flange standard
JIS B 2210 (1984)ANSI B16.5 (1988)
JPI-7S-15-93
3Group 2(ii)
Table 9bar
No limit
No limit
No limit
No limit
No limit
No limit
No limit
No limit
No limit
No limit
No limit
No limit
No limit
20.0
16.6
14.2
12.5
11.1
10.0
10.0
10.0
10.0
10.0
10.0
10.0

MPa
No limit
No limit
No limit
No limit
No limit
No limit
No limit
No limit
No limit
No limit
No limit
No limit
No limit
2.00
1.66
1.42
1.25
1.11
1.00
1.00
1.00
1.00
1.00
1.00
1.00

No. SS2-MGG200-0100

Azbil Corporation

PERFORMANCE SPECIFICATIONS

Optional specifications
Test report

Accuracy

Calibration certificate, withstand voltage test, insulation
resistant, hydrostatic pressure test, physical inspection are
included.

(in combination with the model MGG14C converter)<Size 2.5 to 15 mm (0.1 to 1/2 inch)>
Vs = Velocity of setting range

Traceability certificate

Vs (m/s)

The following three documents are included.
• Traceability System Chart

1.0 ≤ Vs ≤ 10

• Traceability Certificate

0.1 ≤ Vs ≤ 1.0• Test Report

Velocity during
measurement ≥Vs × 40 %Velocity during
measurement ≤Vs × 40 %±0.5 % of rate±(0.1/Vs+0.4)%
of rate±0.2 % of Vs±0.4(0.1/Vs+0.4)%
of Vs<Size 25 to 600 mm (1 to 24 inches)>
Vs = Velocity of setting range

Material certificate
Material certificate for electrode/grounding ring

Gasket for plastic piping

Vs (m/s)

When the detector is being mounted on plastic pipe,
attach this gasket between the lining and the grounding
ring, and between the grounding ring and the plastic pipe
flange.

1.0 ≤ Vs ≤ 10
0.1 ≤ Vs ≤ 1.0

Velocity during
measurement ≥ Vs
×20 %±0.5 % of rate±(0.1/Vs+0.4)%
of rateVelocity during
measurement ≤ Vs
×20 %±0.1 % of Vs±0.2(0.1/Vs+0.4)%
of Vs<Size 700 to 1100 mm (28 to 44 inches)>
Vs = Velocity of setting range

Attaching the tag number to the terminal box
Stamp the tag with the specified number and attach to theterminal box. The maximum number of characters of the
tag number is 8.

Vs (m/s)

Attaching the tag number to the neck section

1.0 ≤ Vs ≤ 10

Stamp the tag with the specified number and attach to theneck section of the detector with stainless wire. The maximum number of characters of the tag number is 16.

0.1 ≤ Vs ≤ 1.0

Velocity during
measurement ≥Vs × 50 %Velocity during
measurement ≤Vs × 50 %±1.0 % of rate±(0.2/Vs+0.8)%
of rate±0.5 % of Vs(0.1/Vs+0.4)%
of Vs

Accuracy is guaranteed by the totalized flow volume under
the condition of continuous flow measurement for 30 seconds or longer.

Water free treatment
Condensation is removed from wetted surfaces.

Oil free treatment
When removed from wetted surfaces.
Note) For additional specifications, please contact your Azbil Corporation representative.

4

Azbil Corporation

No. SS2-MGG200-0100

Additional accuracy:

Measurable fluid pressure range (depending
on Frange rating):

Effect of ambient magnetic field: ±0.2 % FS (at 400 A/m) or
less

PFA/ETFE lining;		-0.098 to +2.94 MPa
Polyurethane rubber lining;		-0.098 to +2.94 MPa
Chloroprene rubber lining;Vibration effectIntegral style: 4.9 m/s2 (0.5 G) max.
Remote style converter: 4.9 m/s2 (0.5 G) max.
Remote style detector: 19.6 m/s2 (2 G) max.

		Integral/remote models

Output fluctuation:
When 1 ≤ Vs ≤ 10 m/s: ±0.1 % FS or less
When 0.1 ≤ Vs ≤ 1 m/s: ±0.1/Vs % FS or less

PFA lining
Diameter 2.5 to 10 mm
2.94 MPa

Measurable fluid temperature range:

1.96 MPa

PFA lining
Diameter(mm)
2.5 to 10
15 to 200
250 to 600

Measurable fluid temperature (ºC)
Integral model Remote model Submersiblemodel−40 to +120
−40 to +100−
−40 to +120
−40 to +160
−40 to +60−40 to +120
−40 to +120
−40 to +60−0.098 MPa−40

Note: The maximum measurable fluid temperature for the submersible model (MGG12) is 60 ºC.

80 to 200
250 to 600

120 °C2.94 MPa1.96 MPa

Measurable fluid temperature (ºC)
Integral model Remote model Submersiblemodel−40 to +120
−40 to +120
−40 to +60−40 to +120
−40 to +120
−40 to +60

0.98 MPa−0.098 MPa−40

Polyurethane rubber lining
Diameter(mm)
25 to 200

100

Diameter 15 to 200 mm

ETFE lining
Diameter(mm)-0.098 to +0.98 MPa

Measurable fluid temperature (ºC)
Integral/remote/submersible models−40 to +50

80 100 120

160 °C

Diameter 250 to 600 mm
1.96 MPa

Chloroprene rubber lining
Diameter(mm)
250 to 600
700 to 1100

0.98 MPa

Measurable fluid temperature (ºC)
Integral/remote models Submersible model−10 to +70−10 to +60−10 to +70−−0.098 MPa
−40

−10

70

100

120 °C: Special support(Please contact us.)5No. SS2-MGG200-0100

Azbil Corporation

Integral/remote models

Submersible model

ETFE lining

PFA lining

Diameter 80 to 200 mm

Diameter 15 to 200 mm

2.94 MPa2.94 MPa1.96 MPa1.96 MPa

0.98 MPa

0.98 MPa−0.098 MPa−40

80 100-0.098 MPa-40

120 °CETFE lining

60

120 °C60

120 °C60

120 °CETFE lining

Diameter 250 to 600 mm

Diameter 80 to 200 mm

1.96 MPa2.94 MPa
0.98 MPa1.96 MPa
0.98 MPa−0.098 MPa
−40

100

120 °C-0.098 MPa-40

Polyurethane rubber lining

PFA/ETFE lining

Diameter 25 to 200 mm

Diameter 250 to 600 mm

2.94 MPa1.96 MPa1.96 MPa

0.98 MPa

0.98 MPa-0.098 MPa-40−0.098 MPa−40

50

120 °C

Chloroprene rubber lining

Polyurethane rubber lining

Diameter 250 to 1100 mm

Diameter 25 to 200 mm

1.96 MPa2.94 MPa

0.98 MPa−0.098 MPa−40

1.96 MPa
0.98 MPa−1070

120 °C-0.098 MPa-40

50

120 °C

Chloroprene rubber lining
Diameter 250 to 600 mm
1.96 MPa
0.98 MPa-0.098 MPa
-40

6

-10

60

120 °CAzbil Corporation

No. SS2-MGG200-0100

Measurable electrical conductivity
Combined with model MGG14C converter 3 μS/cm or more

Measurement flow range
Refer to the minimum/maximum set ranges shown in the table below
Size

Minimum flow velocity range is
0 to 0.1 m/s (0 to 0.33 ft/s)
Minimum range

Maximum flow velocity range is
0 to 10 m/s (0 to 32.8 ft/s)
Maximum range

Conversion
factor K

mm

inch

m3/h

GPM

m3/h

GPM

2.5

0.1

0 to 0.001768

0 to 0.007782

0 to 0.1767

0 to 0.7781

56.59

5

0.2

0 to 0.007069

0 to 0.03113

0 to 0.7068

0 to 3.112

14.15

10

3/8

0 to 0.02828

0 to 0.1246

0 to 2.827

0 to 12.45

3.537

15

1/2

0 to 0.06362

0 to 0.2802

0 to 6.361

0 to 28.01

1.572

25

1

0 to 0.1768

0 to 0.7782

0 to 17.67

0 to 77.81

0.565940

1½

0 to 0.4524

0 to 1.993

0 to 45.23

0 to 199.2

0.221050

20 to 0.7069

0 to 3.113

0 to 70.68

0 to 311.2

0.141565

2½

0 to 1.195

0 to 5.261

0 to 119.4

0 to 526.0

0.0837180

30 to 1.810

0 to 7.969

0 to 180.9

0 to 796.8

0.05526100

40 to 2.828

0 to 12.46

0 to 282.7

0 to 1245

0.03537125

50 to 4.418

0 to 19.46

0 to 441.7

0 to 1945

0.02264150

60 to 6.362

0 to 28.02

0 to636.1

0 to 2801

0.01572200

80 to 11.31

0 to 49.81

0 to 1130

0 to 4980

0.008842

250

100 to 17.68

0 to 77.82

0 to 1767

0 to 7781

0.005659300

120 to 25.45

0 to 112.1

0 to 2544

0 to 11205

0.003930350

140 to 34.64

0 to 152.6

0 to 3463

0 to 15251

0.002887400

16

0 to 45.24

0 to 199.3

0 to 4523

0 to 19920

0.002210450

180 to 57.26

0 to 252.2

0 to 5725

0 to 25211

0.001747

500

20

0 to 70.69

0 to 311.3

0 to 7068

0 to 31125

0.001415600

24

0 to 101.8

0 to 448.3

0 to 10178

0 to 44820

0.0009824

700

28

0 to 138.6

0 to 610.1

0 to 13854

0 to 61005

0.0007218

800

32

0 to 181.0

0 to 796.9

0 to 18095

0 to 79680

0.0005526

900

36

0 to 229.1

0 to 1009

0 to 22902

0 to 100846

0.0004366

1000

40

0 to 282.8

0 to 1246

0 to 28274

0 to 124501

0.0003537

1100

44

0 to 342.2

0 to 1507

0 to 34211

0 to 150646

0.0002923

Flow conversion Velocity V(m/s) = K × Q

1
4- × ---------K = Flow conversion factor = ----------2					Q = Flow rate (m3/h)

7

3600

πD

No. SS2-MGG200-0100

Azbil Corporation

PHYSICAL SPECIFICATIONS

Process wetted material
LiningMain body material

PFA (size 2.5 to 600 mm (0.1 to 24 inches))
ETFE (size 80 to 600 mm (3 to 24 inches))

Measuring pipe materials
SUS304 stainless steel

Polyurethane rubber (size 25 to 200 mm (1 to 8 inches))
Chloroprene rubber (size 250 to 1100 mm (10 to
44 inches))

FlangeSUS304 stainless steel (size 2.5 to 65 mm (0.1 to
2½ inches))

Electrode

Carbon steel + corrosion-preventive coating (size 80 to 600
mm (3 to 24 inches))
Carbon steel (size 700 to 1100 mm ( 28 to 44 inches))

SUS316L, ASTM B574 (Hastelloy C-276 equivalent), titanium, zirconium, tantalum, tungsten-carbide, platinum/
iridium

Case

Grounding ring

SCS13 stainless steel (size 2.5 to 15 mm (0.1 to 1/2 inch))
SUS304 stainless steel (size 25 to 200 mm (1 to 8 inches))
SS400 carbon steel (size 250 to 1100 mm (10 to 44 inches))

SUS316, ASTM B575 (Hastelloy C-276 equivalent), titanium, zirconium, tantalum, platinum

Terminal box

SUS316 (size 2.5 to 15 mm (0.1 to 1/2 inch))

Union joint

Aluminum alloy (remote model)

HosefinishSUS316 (size 2.5 to 15 mm (0.1 to 1/2 inch))Paint

IDF Clamp

Model MGG18

SUS316 (size 2.5 to 15 mm (0.1 to 1/2 inch))Standard

Tri Clamp

Terminal box

SUS316 (size 2.5 to 15 mm (0.1 to 1/2 inch))

Baked acrylic paintGasketDetector case (size 250 to 1100 mm (10 to 44 inches))

PTFE (if the grounding ring is not made of SUS316)

Epoxy paintO-ringCorrosion-resistant paintViton rubber (with union joints)

Terminal box
Baked acrylic paint
Detector case (size 250 to 1100 mm (10 to 44 inches))
Epoxy paintCorrosion-proof paintTerminal box
Epoxy paint
Detector case (size 250 to 1100 mm (10 to 44 inches))
Epoxy paint

Model MGG19
Tar epoxy paint

Color
Model MGG18
Cover: light beige (Munsell 4Y7.2/1.3)
Housing: dark beige (Munsell 10YR4.7/0/5)

Model MGG19
black
8

Azbil Corporation

No. SS2-MGG200-0100

INSTALLATION

Installation location

Ambient temperature limits

Install this product in a place where the inside of the detector
will always be filled with the process fluid. An installation
example is shown in the figure below.

-25 to + 60 °C (-13 to + 140 °F) (integral model)-30 to + 80 °C (-22 to + 176 °F) (remote model, PFA lining)

Air can easily collect-30 to + 60 °C (-22 to + 140 °F) (remote model, polyurethane

May not be
filled with Good
fluid

rubber lining/chloroprene rubber lining)-30 to + 60 °C (-22 to + 140 °F) (Submersible model, PFA/ETFA lining)

May not be
filled with
fluid

Pump
Good

-30 to + 50 °C (-22 to + 122 °F) (Submersible model, polyurethane rubber lining)

Installation example
Note:

Ambient humidity limits• Install the detector in a place like those circled in the above figure so

5 to 100 % RH

that it stays full of fluid. If the detector is used when it is not full of
fluid, an output error may result.

• If the process fluid is highly viscous, installing the detector in a ver-Electrical connection

tical pipe is recommended in order to ensure axisymmetric flow.

Integral model• Provide a straight pipe section upstream of the installation location.

Connected to converter

Refer to the figure below for the straight pipe length.

Remote model
Upstream side

General modelG1/2 (PF1/2) internal thread, 1/2 NPT internal thread,
CM20 internal thread,Pg 13.5 internal thread.

Right-angle joint

FM Nonincendive model

detector

Diffuser with cone angle
greater than 15(If cone angle is 15 or less,considered as straight pipe)

detector

Greater than 5 dia.

1/2NPT internal thread for model MGG18
Watertight gland for model MGG19

Greater than 10 dia.
detector

T joint

Pipe connection
Wafer (size 2.5 to 200 mm (0.1 to 8 inches))
Flange (size 2.5 to 1100 mm (0.1 to 44 inches))
Union (size 2.5 to 15 mm (0.1 to 1/2 inch))
Hose (size 2.5 to 15 mm (0.1 to 1/2 inch))
IDF Clamp (size 2.5 to 15 mm (0.1 to 1/2 inch))
Tri Clamp (size 2.5 to 15 mm (0.1 to 1/2 inch))

Greater than 5 dia.
Gate Value(completely open)

Nuts and bolts (for wafer models)detector

Greater than 5 dia.

S20C carbon steel, SUS304 stainless steel

Any pump

GroundingP

Resistance less than 100 ΩGreater than 10 dia.

Length of straight pipe
Upstream side

Figure 1.

A minimum five straight pipe diameters
A minimum 10 straight pipe diameters is required if a
diffuser/valve/pump is installed upstream side.

Downstream side
Two straight pipe diameters is recommended.

9

Concentrator(considered asstraight-pipe section)

detector

Greater than 5 dia.
Any Control Valuedetector

Greater than 10 dia.

No. SS2-MGG200-0100

Azbil Corporation

Cable (between remote detector andconverter)
Maximum length 300 m (984 ft)(depends on fluid conductivity)
Outer diameter 10 to 12 mm (0.4 to 0.47 inch)

Signal cable
Dedicated cable: MGA12W(O.D. 11.4 mm, 0.75 mm2) or equivalent (CVVS,CEEV etc.)

Excitation cable
Dedicated cable: MGA12W(O.D. 10.5 mm, 2 mm2) or equivalent (CVV and others)

Maximum cable length of MGA12W cable
300
200

Cable length [m]

100
50
20
10
5
2
1
3

5

10

20

50 100200

500 1000

Process fluid conductivity [µ S/cm]

Figure 2. Maximum cable length of MGA12W cable10Azbil Corporation

No. SS2-MGG200-0100

Notice for installation(e.g. rinse, shampoo and CWM (coal water mixture))

To fully enjoy the performance of the device, please choose
an appropriate location according to the following.

• Insulating adhesive materials

Notice after installation• Insufficiently mixed fluid(eg. kaolinite, kaolin, calcium stearate)(Ex.: Fluid just after chemical dosing)

WARNING• If the fluid is cold water and there is a possibility ofWhen removing the device from the piping, make sure that
there is no line pressure or process fluid inside of the device. Removing the device before depressurizing may result
in serious injury.

condensation, select optional specification 6, “Condensation countermeasure,” when ordering.

• The following fluids will permeate the PFA liner. The venthole option is recommended for the following fluids.
• Nitric acid

CAUTION• Aqueous ammonia

Do not use the device as a foothold. It may cause injury or
damage of the device.

• High temperature sodium hydrate

If an electromagnetic flowmeter is installed in air-conditioning equipment, etc., where black pipes are often used for
closed piping and water temperature is about 85 °C, blackrust (a conductive substance) may be generated due to pipe
corrosion. If it sticks to the inner surface of the flowmeter,
the measured output value may drop. To be precise, depending on various environmental conditions such as the amountof dissolved oxygen, black rust may occur even at temperatures around 60 °C. The rate of progress of corrosion, thetype and amount of corrosion, and the amount of adhesion
also differ depending on the environment at the installationsite. If the electromagnetic flowmeter is used in such an
installation environment, it is necessary to control the water
quality to prevent pipe corrosion by measures such as using a
corrosion inhibitor.

Notice for environment• Install the flowmeter in a location with an ambient

temperature of -25 °C to 60 °C (-13 °F to 140 °F) and an
ambient humidity of 5 to 100%RH to prevent equipment
malfunction or output errors.

• Do not install the flowmeter near high-current power lines,
motors or transformers to prevent damage from electromagnetic induction, which can cause equipment malfunction or output errors.

• Do not install the flowmeter in a location subject to severe
vibration or in a highly corrosive atmosphere. The converter and detector can be damaged.

• When install some electromagnetic flowmeters in closer

To further ensure reliable measurement, periodic wiping of
the inside of the electromagnetic flowmeter is needed.

location, keep minimum 500 mm (20 inch) space fromeach flowmeter. Closer electromagnetic flowmeter installation may cause magnetic interference each other andresults in output errors.

* Please contact an Azbil representative for cleaning of theinside of the electromagnetic flowmeter.

• When installing DC-powered electromagnetic flow meters
adjacent to each other, make sure that there is a space of
500 mm or more between the ends of the detectors.

Notes on installation location:• Legs are attached to some models to prevent them from

Notice for application• Electrochemically homogeneous fluid

falling over before installation. If the product is installedwith the legs attached, please also consider earthquake
resistance where appropriate.

Install the device where the process fluid is electrochemically homogeneous. If two kind of process fluids are mixed
at the upstream side, the process fluid must be uniformly
mixed.

Notes on submersible models:• The entire surface of the detector’s terminal box is covered
with waterproof paint. If opened, the terminal box is nolonger waterproof.

• The application which the electric conductivity changes or

non-homogeneous fluid
Do not use the device for the following fluid conditions
even if the electric conductivity, temperature, and pressure
are within the device specifications. Those fluid may cause
of inaccurate flow measurement.
• Fluids that have sufficient conductivity at high tem-

perature but do not meet the conductivity requirement
at room temperature (about 20 °C (68 °F)).
(e.g. fatty acids and soap)• Some fluids contain surfactant
11No. SS2-MGG200-0100

Azbil Corporation

For FM Nonincendive modelThis equipment is suitable for use in Class I, Division 2, Groups A, B, C and D, Class II, Division 2, Groups F and G; Class III,
Division 2.
If the combination of detector MGG 18/19 and converter MGG 14 C is used as an FM - NI product, both the detector and theconverter must be used in combination with the NI specification.

CAUTION
Power supply and internal voltage of ordinary equipment to the earth shall not exceed 250 V AC 50/60 Hz, 250 V DC in caseof normal /fault conditions.

Table 2TYPE

MODEL NO. MAX.AMBIENT TEMP. MAX.FLUID TEMP

MGG14CMGM14CMGM18D,F
INTEGRAL
MGG18D,F,U
MGG18D
MGS18U
MGG14CREMOTENote––
120 °C
120 °C
50 °C
120 °C–
160 °C

60 °C

60 °C

MGG18D,F

80 °C

MGG18D
MGG18U
MGG19D,F,U
MGG19D
MGS18U

60 °C

120 °C
50 °C
120 °C
120 °C
50 °C
160 °C

80 °C
60 °C
80 °C

LINING

1. Ambient Temperature, Process Temperature: See table 2.
2. Power Supply and Internal Voltage of Ordinary Equipment to the Earth.
shall not exceed AC250V 50/60Hz, DC250V incase of Normal/Fault conditions.
3. In Division 2 Location.

12

SIZE

–
–
–
–
PFA,ETFE
40 to 600A
PFA,ETFE
2.5 to 600A
POLYURETHANE 25 to 200A
PFA
15 to 125A––
PFA
2.5 to 200A
PFA
250 to 600A
ETFE
2.5 to 600A
POLYURETHANE 25 to 200A
PFA,ETFE
2.5 to 15A
PFA,ETFE
2.5 to 600A
POLYURETHANE 25 to 200A
PFA
15 to 125A

Azbil Corporation

No. SS2-MGG200-0100• Fluid being measured must be non-flammable.
• Install Wiring per NEC 501-4(b) or 502-4(b).
4. Degree of Protetion of Enclosure
MGG14C, MGG18D,U,F, MGS18U, MGM14C, MGM18D,F:

Type 4X

MGG19D,U,F:						Type 6P

MODEL SELECTION
Contents of model number table
Detector (General model)
Structure / Basic model no.

LiningPipe connection

Size

Ref. page

Watertight model MGG18U

PFA

Union / Hose / Clamp

2.5 to 15 mm(0.1 to 1/2 inch)

page 14

Watertight model MGG18D

PFA

Wafer

2.5 to 10 mm(0.1 to 3/8 inch)

page 15

Watertight model MGG18D

PFA / ETFE

Wafer

15 to 200 mm(1/2 to 8 inches)

page 16

Watertight model MGG18F

PFA / ETFE

Flange15 to 200 mm(1/2 to 8 inches)

page 17

Watertight model MGG18F

PFA / ETFE

Flange

250 to 600 mm(10 to 24 inches)

page 18

Watertight model MGG18D

Polyurethane rubber

Wafer

25 to 200 mm(1 to 8 inches)

page 19

Watertight model MGG18F

Chloroprene rubber

Flange

250 to 600 mm(10 to 24 inches)

page 20

Watertight model MGG11F

Chloroprene rubber

Flange700 to 1100 mm (28 to 44 inches)

page 21Structure / Basic model no.

LiningPipe connection

Size

Ref. page

Submersible model MGG19D

PFA / ETFE

Wafer

Submersible model MGG19F

PFA / ETFE

FlangeSubmersible model MGG19F

PFA / ETFE

FlangeSubmersible model MGG19D

Polyurethane rubber

Wafer

25 to 200 mm

Detector (Submersible model)

Note)

15 to 200 mm(1/2 to 8 inches)

page 2215 to 200 mm(1/2 to 8 inches)

page 23

250 to 600 mm(10 to 24 inches)

page 24(1 to 8 inches)

page 25

All MGG19 models satisfy FM Nonincendive approval.
PFA / ETFE lining
Rubber lining

Lining CharacteristicsPFA:
PFA is a chemical-resistant, heat-resistant, and adhesion-resistant lining material that can be used for almost any corrosive
liquid. Select this lining for use with corrosive liquids (sulfuric acid, hydrochloric acid, caustic soda, acetic acid, etc.). However, for nitric acid and hydrofluoric acid, the service life may be shorter if the concentration and pressure are high.

ETFE:
Chemical resistance is slightly lower than that of a PFA lining. Do not use ETFE for strongly corrosive liquids such as sulfuric acid, fluoric acid, nitric acid, and hydrochloric acid. In terms of abrasion resistance, ETFE is about 1.5 times stronger
than PFA. Therefore, it can be used for pulp slurry (except for black liquor) and will have a longer service life than PFA.
However, because it has lower heat resistance than PFA, it cannot be used in a pipeline with fluids at 120 °C or higher. Do
not use the flowmeter in a pipeline that will be cleaned with steam.

Rubber:Both polyurethane and chloroprene are excellent for abrasion resistance, but because they have little chemical resistance,
they cannot be used for corrosive liquids.

13

No. SS2-MGG200-0100

Azbil Corporation

Union / Hose / Clamp type (2.5 to 15 mm (0.1 to 1/2 inch)) PFA lining
Model MGG18U - I II III IV V VI VII VIII IX - X - Y / Options (some options can be selected per each model)
Basic model no.

Selections--

I

Line size

II
III

LiningPipingconnection

IV

Electrode

V
VI

Grounding ringElectricalconnection /watertight gland

VII
VIII

Face-to-faceInstallation /wiring direction

IX

Calibration

X

FinishOptions

MGG18U

Azbil Corporation version (must be selected)
Traceability certificate for detector
Material certificate (only for electrodes and ground rings)
Attachment of the TAG number to the terminal box for detector (Note 2)
Attachment of the TAG number plate to the neck section for detector (Note 2) (Note 3)
Water free treatment
Oil free treatment

Note)

Optional selections

2.5 mm(0.1 inch)002
5 mm(0.2 inch)
005
10 mm
(3/8 inch)
010
15 mm
(1/2 inch)
015
PFA
P
Union joint R1/2 (PT1/2) external thread
U1
Union joint 1/2NPT external thread
U2
Union joint R1/2 (PT1/2) internal thread
U3
Union joint 1/2NPT internal thread
U4
Hose joint
H1
IDF clamp
C1
Tri clamp
C2
SUS316L
ASTM B574 (Hastelloy C-276 equivalent)
Titanium
Zirconium
Tungsten carbide (only for size 10 mm or upper)
Other
SUS316Integral type
G1/2 internal thread / without watertight gland
Remotetype
G1/2 internal thread / with brass (Ni-plated) watertight glandG1/2 internal thread / with plastic watertight gland
1/2NPT internal thread / without watertight gland (Note 1)
CM20 internal thread / without watertight glandPg 13.5 internal thread / without watertight glandG1/2 internal thread / with SUS304 watertight glandStandard
Integral type
Upstream side (horizontal / vertical piping mounting)
Remotetype
Downstream side (horizontal / vertical piping mounting)
Horizontal piping mounting / left side viewed from upstream
Horizontal piping mounting / right side viewed from upstream
Standard
Other
Standard
Corrosion-resistant finish
Corrosion-proof finish

L
C
K
H
W
_
S
1
2
3
4
5
6
7
8
A
H
A
B
C
D
A
_
X
1
2

1. Must be selected for FM NI approval			

2. Must be selected for Tag no. requirement			

3. This option code cannot be selected with Electrical connection code “5”.

14

Y
B
C
K
L
E
F

Azbil Corporation

No. SS2-MGG200-0100

Wafer type (2.5 to 10 mm (0.1 to 3/8 inch)) PFA lining
Model MGG18D - I II III IV V VI VII VIII IX - X XI - Y / Options (some options can be selected per each model)
Basic model no.

Selections

MGG18D--I

Line size

II
III

LiningPipingconnection

IV

ElectrodeV

Grounding ring

SUS316
ASTM B575 (Hastelloy C-276 equivalent)
Titanium
Zirconium
Tantalum
Platinum
Other

VI

Electricalconnection /watertight glandVII

Face-to-facedimensions
Installation /wiring direction

Integral type
G1/2 internal thread / without watertight gland
RemoteG1/2 internal thread / with brass (Ni-plated) watertight gland
type
G1/2 internal thread / with plastic watertight gland
1/2NPT internal thread / without watertight gland (Note 1)
CM20 internal thread / without watertight glandPg 13.5 internal thread / without watertight glandG1/2 internal thread / with SUS304 watertight glandStandard
Other

2.5 mm(0.1 inch)
5 mm(0.2 inch)
10 mm(3/8 inch)
PFA
Wafer JIS 10K
Wafer JIS 20K
Wafer JIS 30K
Wafer JIS 10/20K for 10 mm flange
Wafer JIS 30K for 10 mm flange
Wafer ANSI 150
Wafer ANSI 300
Wafer DIN PN10
Wafer DIN PN16
Wafer DIN PN25
Wafer DIN PN40
Wafer DIN PN10/16/25/40 for 10 mm flange
Wafer JPI 150
Wafer JPI 300
SUS316L
ASTM B574 (Hastelloy C-276 equivalent)
Titanium
Zirconium
Tantalum
Tungsten carbide (only for size 10 mm)
Platinum iridium
Other

002
005
010
P
11
12
13
14
15
21
22
41
42
43
44
45
61
62
L
C
K
H
T
W
P

_
S
C
K
H
T
P

_

Integral type
Upstream side (horizontal / vertical piping mounting)
RemoteDownstream side (horizontal / vertical piping mounting)type
Horizontal piping mounting / left side viewed from upstream
Horizontal piping mounting / right side viewed from upstream
Standard
Other

IX

Calibration

X

FinishXI

Bolt / nut

Options

VIII

Optional selections

Azbil Corporation version (must be selected)
Traceability certificate for detector
Material certificate (only for electrodes and ground rings)
With gasket for plastic piping
Attachment of the TAG number to the terminal box for detector (Note 2)
Attachment of the TAG number plate to the neck section for detector (Note 2) (Note 3)

Y
B
C
J
K
L

Water free treatment

E

Oil free treatment

F

Standard
Corrosion-resistant finish
Corrosion-proof finish
None
Carbon steel
SUS304

1
2
3
4
5
6
7
8
A

_
H
A
B
C
D
A

_
X
1
2
X
1
2

15

Note) 1. Must be selected for FM NI approval
2. Must be selected for Tag no. requirement
3. This option code cannot be selected with
Electrical connection code “5”.

No. SS2-MGG200-0100

Azbil Corporation

Wafer type (15 to 200 mm (1/2 to 8 inches)) PFA / ETFE lining
Model MGG18D - I II III IV V VI VII VIII IX - X XI - Y / Options (some options can be selected per each model)
Basic model no.

Selections

MGG18D-I

Line size

II

LiningIII

Pipingconnection

IV

ElectrodeV

Grounding ring

SUS316
ASTM B575 (Hastelloy C-276 equivalent)
Titanium
Zirconium
Tantalum
Platinum
Other

VI

Electricalconnection /watertight gland

Integral type
Remote typeVII

Face-to-facedimensions
Installation /wiring direction

Standard
Other
Integral type
Remote type

Calibration

Standard+/- 0.35 % of rate calibration (Size 40 to 200 mm (1 1/2 to 8 inches))
Other

VIII

15 mm
(1/2 inch)
25 mm(1 inch)40 mm(1½ inches)
50 mm(2 inches)
65 mm(2½ inches)
80 mm(3 inches)
100 mm(4 inches)
125 mm(5 inches)
150 mm(6 inches)
200 mm(8 inches)
ETFE (Size 80 to 200 mm (3 to 8 inches))
PFA
Wafer JIS 10K
Wafer JIS 20K
Wafer JIS 30K
Wafer ANSI 150
Wafer ANSI 300
Wafer JIS G3443-2 F12 (size 80 mm or larger)
Wafer DIN PN10
Wafer DIN PN16
Wafer DIN PN25
Wafer DIN PN40
Wafer JPI 150
Wafer JPI 300
SUS316L
ASTM B574 (Hastelloy C-276 equivalent)
Titanium
Zirconium
Tantalum
Tungsten carbide
Platinum iridium
Other

015
025
040
050
065
080
100
125
150
200
E
P
11
12
13
21
22
31
41
42
43
44
61
62
L
C
K
H
T
W
P

_

S
C
K
H
T
P

_

G1/2 internal thread / without watertight glandG1/2 internal thread / with brass (Ni-plated) watertight glandG1/2 internal thread / with plastic watertight gland
1/2NPT internal thread / without watertight gland (Note 1)
CM20 internal thread / without watertight glandPg 13.5 internal thread / without watertight glandG1/2 internal thread / with SUS304 watertight gland

1
2
3
4
5
6
7
8
A

_
Upstream side (horizontal / vertical piping mounting)Downstream side (horizontal / vertical piping mounting)
Horizontal piping mounting / left side viewed from upstream
Horizontal piping mounting / right side viewed from upstream

X

FinishXI

Bolt / nut

Options

IX

Optional selections-

Azbil Corporation version (must be selected)
Traceability certificate for detector
Material certificate (only for electrodes and ground rings)
With gasket for plastic piping
Attachment of the TAG number to the terminal box for detector (Note 2)
Attachment of the TAG number plate to the neck section for detector (Note 2) (Note 3)
Water free treatment
Oil free treatment

H
A
B
C
D
A
U

_

Standard
Corrosion-resistant finish
Corrosion-proof finish
None
Carbon steel
SUS304

X
1
2
X
1
2

16

Y
B
C
J
K
L
E
F

Note) 1. Must be selected for FM NI approval
2. Must be selected for Tag no. requirement
3. This option code cannot be selected with
Electrical connection code “5”.

Azbil Corporation

No. SS2-MGG200-0100

Flange type (15 to 200 mm (1/2 to 8 inches)) PFA / ETFE lining
Model MGG18F - I II III IV V VI VII VIII IX X - XI - Y / Options (some options can be selected per each model)
Basic model no.

Selections

MGG18F

I

Line size

II

LiningIII

Pipingconnection

IV

Flange material

V

Electrode--(1/2 inch)
15 mm(1 inch)25 mm(1½ inches)
40 mm(2 inches)
50 mm(2½ inches)
65 mm(3 inches)
80 mm(4 inches)
100 mm(5 inches)
125 mm(6 inches)
150 mm(8 inches)
200 mmETFE (Size 80 to 200 mm (3 to 8 inches))
PFA
Flange JIS 10K
Flange JIS 20K
Flange JIS 30K
Flange ANSI 150
Flange ANSI 300
Flange JIS G3443-2 F12 (line size 80 mm or larger)
Flange DIN PN10
Flange DIN PN16
Flange DIN PN25
Flange DIN PN40
Flange JPI 150
Flange JPI 300
Standard
Other

015
025
040
050
065
080
100
125
150
200
E
P
J1
J2
J3
A1
A2
G1
D1
D2
D3
D4
P1
P2
1

_

SUS316L
ASTM B574 (Hastelloy C-276 equivalent)

L
C
K
H
T
W
P

Titanium
Zirconium
Tantalum
Tungsten carbide
Platinum iridium
Other

VI

Grounding ring

Optional selections_

SUS316
ASTM B575 (Hastelloy C-276 equivalent)

S
C
K
H
T
P

Titanium
Zirconium
Tantalum
Platinum
Other

_

VII

Electricalconnection /watertight gland

Integral type
Remote type

VIII

Face-to-facedimensions
Installation /wiring direction

Standard

A

Other
Integral type_

Calibration

Standard+/- 0.35 % of rate calibration (Size 40 to 200 mm (1 1/2 to 8 inches))
Other

IX

Upstream side (horizontal / vertical piping mounting)Downstream side (horizontal / vertical piping mounting)
Horizontal piping mounting / left side viewed from upstream
Horizontal piping mounting / right side viewed from upstream

XI

FinishOptions

X

Remote type

G1/2 internal thread / without watertight glandG1/2 internal thread / with brass (Ni-plated) watertight glandG1/2 internal thread / with plastic watertight gland
1/2NPT internal thread / without watertight gland (Note 1)
CM20 internal thread / without watertight glandPg 13.5 internal thread / without watertight glandG1/2 internal thread / with SUS304 watertight gland

Azbil Corporation version (must be selected)
Traceability certificate for detector
Material certificate (only for electrodes and ground rings)
With gasket for plastic piping

1
2
3
4
5
6
7
8

H
A
B
C
D
A
U

_

Standard
Corrosion-resistant finish
Corrosion-proof finish

X
1
2

Attachment of the TAG number to the terminal box for detector (Note 2)
Attachment of the TAG number plate to the neck section for detector (Note 2) (Note 3)
Water free treatment
Oil free treatment

17

Y
B
C
J
K
L
E
F

Note) 1. Must be selected for FM NI approval
2. Must be selected for Tag no. requirement
3. This option code cannot be selected with
Electrical connection code “5”.

No. SS2-MGG200-0100

Azbil Corporation

Flange type (250 to 600 mm (10 to 24 inches)) PFA / ETFE lining
Model MGG18F - I II III IV V VI VII VIII IX X - XI - Y / Options (some options can be selected per each model)
Basic model no.

Selections

MGG18F

I

Line size

II

LiningIII

Pipingconnection

IV

Flange material

V

Electrode--(10 inches)
250 mm(12 inches)
300 mm(14 inches)
350 mm(16 inches)
400 mm(18 inches)
450 mm(20 inches)
500 mm(24 inches)
600 mmETFE
PFAFlange JIS 10K
Flange JIS 20K
Flange ANSI 150
Flange ANSI 300 (Size 16 inches or smaller)
Flange JIS G3443-2 F12Flange DIN PN10
Flange DIN PN16
Flange DIN PN25
Flange JPI 150
Flange JPI 300 (Size 400 mm or smaller)
Standard
Other

250
300
350
400
450
500
600
E
P
J1
J2
A1
A2
G1
D1
D2
D3
P1
P2
1

_

SUS316L
ASTM B574 (Hastelloy C-276 equivalent)

L
C
K
H
T
W
P

Titanium
Zirconium
Tantalum
Tungsten carbide
Platinum iridium
Other

VI

Grounding ring

Optional selections_

SUS316
ASTM B575 (Hastelloy C-276 equivalent)

S
CK

Titanium
Other

_

VII

Electricalconnection /watertight gland

Integral type
Remote type

VIII

Face-to-facedimensions
Installation /wiring direction

Standard

A

Other
Integral type_

Calibration

Standard+/- 0.35 % of rate calibration (Size 250 to 350 mm (10 to 14 inches))
Other

IX

Upstream side (horizontal / vertical piping mounting)Downstream side (horizontal / vertical piping mounting)
Horizontal piping mounting / left side viewed from upstream
Horizontal piping mounting / right side viewed from upstream

XI

FinishOptions

X

Remote type

G1/2 internal thread / without watertight glandG1/2 internal thread / with brass (Ni-plated) watertight glandG1/2 internal thread / with plastic watertight gland
1/2NPT internal thread / without watertight gland (Note 1)
CM20 internal thread / without watertight glandPg 13.5 internal thread / without watertight glandG1/2 internal thread / with SUS304 watertight gland

Azbil Corporation version (must be selected)
Traceability certificate for detector
Material certificate (only for electrodes and ground rings)
With gasket for plastic piping

1
2
3
4
5
6
7
8

H
A
B
C
D
A
U

_

Standard
Corrosion-resistant finish
Corrosion-proof finish

X
1
2

Attachment of the TAG number to the terminal box for detector (Note 2)
Attachment of the TAG number plate to the neck section for detector (Note 2) (Note 3)
Water free treatment
Oil free treatment

18

Y
B
C
J
K
L
E
F

Note) 1. Must be selected for FM NI approval
2. Must be selected for Tag no. requirement
3. This option code cannot be selected with
Electrical connection code “5”.

Azbil Corporation

No. SS2-MGG200-0100

Wafer type (25 to 200 mm (1 to 8 inches)) Polyurethane rubber lining
Model MGG18D - I II III IV V VI VII VIII IX - X XI - Y / Options (some options can be selected per each model)
Basic model no.

Selections

MGG18D-Line size

25 mm
40 mm
50 mm
65 mm
80 mm
100 mm
125 mm
150 mm
200 mmII
III

LiningPipingconnection

Polyurethane rubber

IV

Electrode

SUS316L
Titanium
Tungsten carbide (only for size 10 mm)
Other

V

Grounding ring

SUS316
Titanium
Other

VI

Electricalconnection /watertight gland

Integral type
Remote typeVII

Face-to-facedimensions
Installation /wiring direction

Calibration

025
040
050
065
080
100
125
150
200
Q

Wafer JIS 10K
Wafer JIS 20K
Wafer JIS 30K
Wafer ANSI 150
Wafer ANSI 300
Wafer JIS G3443-2 F12 (line size 80 mm or larger)
Wafer DIN PN10
Wafer DIN PN16
Wafer DIN PN25
Wafer DIN PN40
Wafer JPI 150
Wafer JPI 300

11
12
13
21
22
31
41
42
43
44
61
62
L
K
W

_
S
K

_
G1/2 internal thread / without watertight glandG1/2 internal thread / with brass (Ni-plated) watertight glandG1/2 internal thread / with plastic watertight gland
1/2NPT internal thread / without watertight gland (Note 1)
CM20 internal thread / without watertight glandPg 13.5 internal thread / without watertight glandG1/2 internal thread / with SUS304 watertight glandStandard
Remote type

1
2
3
4
5
6
7
8
A

Integral type
Upstream side (horizontal / vertical piping mounting)Downstream side (horizontal / vertical piping mounting)
Horizontal piping mounting / left side viewed from upstream
Horizontal piping mounting / right side viewed from upstream

Standard+/- 0.35 % of rate calibration (Size 40 to 200 mm (1 1/2 to 8 inches))
Other

X

FinishStandard
Corrosion-resistant finish
Corrosion-proof finish

XI

Bolt / nut

None
Carbon steel
SUS304

Options

IX-(1 inch)(1½ inches)(2 inches)(2½ inches)(3 inches)(4 inches)(5 inches)(6 inches)(8 inches)

I

VIII

Optional selections

Azbil Corporation version (must be selected)
Traceability certificate for detector
Material certificate (only for electrodes and ground rings)
With gasket for plastic piping

H
A
B
C
D
A
U

_
X
1
2
X
1
2

Attachment of the TAG number to the terminal box for detector (Note 2)
Attachment of the TAG number plate to the neck section for detector (Note 2) (Note 3)
Water free treatment
Oil free treatment

19

Y
B
C
J
K
L
E
F

Note) 1. Must be selected for FM NI approval
2. Must be selected for Tag no. requirement
3. This option code cannot be selected with
Electrical connection code “5”.

No. SS2-MGG200-0100

Azbil Corporation

Flange type (250 to 600 mm (10 to 24 inches)) Chloroprene rubber lining
Model MGG18F - I II III IV V VI VII VIII IX X - XI - Y / Options (some options can be selected per each model)
Basic model no.

Selections

MGG18F--

Line size

II
III

LiningPipingconnection

IV

Flange material

V

ElectrodeVI

Grounding ringVII

Electricalconnection /watertight gland

VIII

Face-to-facedimensions

IX

Installation /wiring direction

X

Calibration

XI

FinishOptions(10 inches)
250 mm250(12 inches)
300 mm300(14 inches)
350 mm350(16 inches)
400 mm400(18 inches)
450 mm450(20 inches)
500 mm500(24 inches)
600 mm600
Chloroprene rubber
R
Flange JIS 10K
J1Flange ANSI 150
A1
Flange JIS G3443-2 F12
G1Flange DIN PN10
D1
Flange JPI 150
P1Standard
1
Other_
SUS316L
Titanium
Tungsten carbide
Other
SUS316
Titanium
Other
Integral type
G1/2 internal thread / without watertight gland
Remotetype
G1/2 internal thread / with brass (Ni-plated) watertight glandG1/2 internal thread / with plastic watertight gland
1/2NPT internal thread / without watertight gland (Note 1)
CM20 internal thread / without watertight glandPg 13.5 internal thread / without watertight glandG1/2 internal thread / with SUS304 watertight glandStandard
Other
Integral type
Upstream side (horizontal / vertical piping mounting)
Remotetype
Downstream side (horizontal / vertical piping mounting)
Horizontal piping mounting / left side viewed from upstream
Horizontal piping mounting / right side viewed from upstream
Standard+/- 0.35 % of rate calibration (Size 250 to 350 mm (10 to 14 inches))
Other

I

Azbil Corporation version (must be selected)
Traceability certificate for detector
Material certificate (only for electrodes and ground rings)
With gasket for plastic piping
Attachment of the TAG number to the terminal box for detector (Note 2)
Attachment of the TAG number plate to the neck section for detector (Note 2) (Note 3)
Water free treatment
Oil free treatment

Note)

Optional selections

Standard
Corrosion-resistant finish
Corrosion-proof finish

L
K
W
_
S
K
_
1
2
3
4
5
6
7
8
A
_
H
A
B
C
D
A
U
_
X
1
2

1. Must be selected for FM NI approval			

2. Must be selected for Tag no. requirement			

3. This option code cannot be selected with Electrical connection code “5”.

20

Y
B
C
J
K
L
E
F

Azbil Corporation

No. SS2-MGG200-0100

Flange type (700 to 1100 mm (28 to 44 inches)) Chloroprene rubber lining
Model MGG11F - I II III IV V VI VII VIII IX X - XI - Y / Options (some options can be selected per each model)
Basic model no.

Selections

MGG11F--

Line size

II
III

LiningPipingconnection

IVV

Flange material
ElectrodeVI

Grounding ringVII

Electricalconnection /watertight gland

VIII

Face-to-facedimensions

IX

Installation /wiring direction

X

Calibration

XI

FinishOptions(28 inches)
700 mm700(32 inches)
800 mm
800(36 inches)
900 mm900(40 inches)1000 mm
10H(44 inches)1100 mm
11H
Chloroprene rubber
R
Wafer JIS 10K
J1
Wafer ANSI 150
A1
Wafer JIS G3443-2 F12G1
Wafer DIN PN10
D1
Wafer JPI 150
P1Standard
1
SUS316L
Titanium
Tungsten carbide
Other
SUS316
Other
Integral type
G1/2 internal thread / without watertight gland
Remotetype
G1/2 internal thread / with brass (Ni-plated) watertight glandG1/2 internal thread / with plastic watertight gland
1/2NPT internal thread / without watertight gland
CM20 internal thread / without watertight glandPg 13.5 internal thread / without watertight glandG1/2 internal thread / with SUS304 watertight glandStandard
Other
Integral type
Upstream side (horizontal / vertical piping mounting)
Remotetype
Downstream side (horizontal / vertical piping mounting)
Horizontal piping mounting / left side viewed from upstream
Horizontal piping mounting / right side viewed from upstream
Standard
Other

I

Azbil Corporation version (must be selected)
Traceability certificate for detector
Material certificate (only for electrodes and ground rings)
Attachment of the TAG number to the terminal box for detector (Note 1)
Attachment of the TAG number plate to the neck section for detector (Note 1)
Water free treatment
Oil free treatment

Note)

Optional selections

Standard
Corrosion-resistant finish
Corrosion-proof finish

L
K
W
_
S
_
1
2
3
4
5
6
7
8
A
_
H
A
B
C
D
A
_
X
1
2

1. Must be selected for Tag no. requirement

21

Y
B
C
K
L
E
F

No. SS2-MGG200-0100

Azbil Corporation

Submersible detector with FM NI approval
Wafer type (15 to 200 mm (1/2 to 8 inches)) PFA / ETFE lining
Model MGG19D - I II III IV V VI VII VIII IX - X XI - Y / Options (some options can be selected per each model)
Basic model no.

Selections

MGG19DI

Line size

II

LiningIII

Piping connection

IV

Electrode

Grounding ring

(1/2 inch)
15 mm(1 inch)25 mm(1½ inches)
40 mm(2 inches)
50 mm(2½ inches)
65 mm(3 inches)
80 mm(4 inches)
100 mm(5 inches)
125 mm(6 inches)
150 mm(8 inches)
200 mmETFE (Size 80 to 200 mm (3 to 8 inches))
PFA
Wafer JIS 10K
Wafer JIS 20K
Wafer JIS 30K
Wafer ANSI 150
Wafer ANSI 300
Wafer JIS G3443-2 F12 (line size 80 mm or larger)
Wafer DIN PN10
Wafer DIN PN16
Wafer DIN PN25
Wafer DIN PN40
Wafer JPI 150
Wafer JPI 300

015
025
040
050
065
080
100
125
150
200
E
P
11
12
13
21
22
31
41
42
43
44
61
62

SUS316L
ASTM B574 (Hastelloy C-276 equivalent)

L
C
K
H
T
W
P

_

SUS316
ASTM B575 (Hastelloy C-276 equivalent)

S
C
K
H
T
P

Titanium
Zirconium
Tantalum
Platinum
Other

VI
VII
VIII

_
G1/2 internal thread / with brass (Ni-plated) watertight gland3

G1/2 internal thread / with SUS304 watertight gland

8

Face-to-facedimensions
Installation / wiring
direction

Standard

Calibration

Standard+/- 0.35 % of rate calibration (Size 40 to 200 mm (1 1/2 to 8 inches))
Other

A

_

Other
Remote type

Upstream side (horizontal / vertical piping mounting)Downstream side (horizontal / vertical piping mounting)
Horizontal piping mounting / left side viewed from upstream
Horizontal piping mounting / right side viewed from upstream

X
XI

FinishBolt / nut

Options

IX

Electrical connection Remote type/ watertight gland

Azbil Corporation version (must be selected)
Traceability certificate for detector
Material certificate (only for electrodes and ground rings)

Note)

selections-

Titanium
Zirconium
Tantalum
Tungsten carbide
Platinum iridium
Other

V

Optional

-

A
B
C
D
A
U

_
X

Standard
None
Carbon steel
SUS304X
1
2

With gasket for plastic piping
Attachment of the TAG number to the terminal box for detector (Note 1)

Y
B
C
J
K

Water free treatment

E

Oil free treatment

F

1. Must be selected for Tag no. requirement

22

Azbil Corporation

No. SS2-MGG200-0100

Submersible detector with FM NI approval
Flange type (15 to 200 mm (1/2 to 8 inches)) PFA / ETFE lining
Model MGG19F - I II III IV V VI VII VIII IX X - XI - Y / Options (some options can be selected per each model)
Basic model no.

Selections

MGG19F

Optional selections--(1/2 inch)
15 mm(1 inch)25 mm(1½ inches)
40 mm(2 inches)
50 mm(2½ inches)
65 mm(3 inches)
80 mm(4 inches)
100 mm(5 inches)
125 mm(6 inches)
150 mm(8 inches)
200 mmETFE (Size 80 to 200 mm (3 to 8 inches))
PFA
Flange JIS 10K
Flange JIS 20K
Flange JIS 30K
Flange ANSI 150
Flange ANSI 300
Flange JIS G3443-2 F12 (line size 80 mm or larger)
Flange DIN PN10
Flange DIN PN16
Flange DIN PN25
Flange DIN PN40
Standard

I

Line size

II

LiningIII

Piping connection

IV

Flange material

V

Electrode

SUS316L
ASTM B574 (Hastelloy C-276 equivalent)
Titanium
Zirconium
Tantalum
Tungsten carbide
Platinum iridium
Other

VI

Grounding ring

SUS316
ASTM B575 (Hastelloy C-276 equivalent)
Titanium
Zirconium
Tantalum
Platinum
Other

VII

Electrical connection Remote type/ watertight gland

VIII

Face-to-facedimensions
Installation / wiring
direction

Standard

Calibration

Standard+/- 0.35 % of rate calibration (Size 40 to 200 mm (1 1/2 to 8 inches))

A
U

Other

_

IX

E
P
J1
J2
J3
A1
A2
G1
D1
D2
D3
D4
1

_

Other

L
C
K
H
T
W
P

_
S
C
K
H
T
P

_

G1/2 internal thread / with brass (Ni-plated) watertight gland3

G1/2 internal thread / with SUS304 watertight gland

8
A

_

Other
Remote type

Upstream side (horizontal / vertical piping mounting)Downstream side (horizontal / vertical piping mounting)
Horizontal piping mounting / left side viewed from upstream
Horizontal piping mounting / right side viewed from upstream

XI

FinishOptions

X

015
025
040
050
065
080
100
125
150
200

Azbil Corporation version (must be selected)
Traceability certificate for detector
Material certificate (only for electrodes and ground rings)

X

Standard
Y
B
C
J
K
E
F

With gasket for plastic piping
Attachment of the TAG number to the terminal box for detector (Note 1)
Water free treatment
Oil free treatment

Note)

A
B
C
D

1. Must be selected for Tag no. requirement23

No. SS2-MGG200-0100

Azbil Corporation

Submersible detector with FM NI approval
Flange type (250 to 600 mm (10 to 24 inches)) PFA / ETFE lining
Model MGG19F - I II III IV V VI VII VIII IX X - XI - Y / Options (some options can be selected per each model)
Basic model no.

Selections

MGG19F
I

II
III

IV
V

VI

VII
VIII
IX

-

(10 inches)
250 mm250(12 inches)
300 mm300(14 inches)
350 mm350(16 inches)
400 mm400(18 inches)
450 mm450(20 inches)
500 mm500(24 inches)
600 mm600
LiningETFE
E
PFA
PFlange JIS 10K
Piping connection
J1
Flange JIS 20K
J2Flange ANSI 150
A1
Flange ANSI 300 (Size 16 inches or smaller)
A2
Flange JIS G3443-2 F12
G1Flange DIN PN10
D1
Flange DIN PN16
D2
Flange DIN PN25
D3
Flange material
Standard
1
Other_
Electrode
SUS316LL
ASTM B574 (Hastelloy C-276 equivalent)C
TitaniumK
Zirconium
H
TantalumT
Tungsten carbide
W
Platinum iridium
P
Other
_
Grounding ring
SUS316
ASTM B575 (Hastelloy C-276 equivalent)
Titanium
Other
G1/2 internal thread / with brass (Ni-plated) watertight glandElectrical connection Remote/ watertight gland
type
G1/2 internal thread / with SUS304 watertight glandFace-to-faceStandard
dimensions
Other
Upstream side (horizontal / vertical piping mounting)Installation / wiring Remote
direction
type
Downstream side (horizontal / vertical piping mounting)
Horizontal piping mounting / left side viewed from upstream
Horizontal piping mounting / right side viewed from upstream
Calibration
Standard+/- 0.35 % of rate calibration (Size 250 to 350 mm (10 to 14 inches))
Other
Line size

XI

FinishOptions

X

Azbil Corporation version (must be selected)
Traceability certificate for detector
Material certificate (only for electrodes and ground rings)

S
C
K
_
3
8
A
_
A
B
C
D
A
U
_
X

Standard
Y
B
C
J
K
E
F

With gasket for plastic piping
Attachment of the TAG number to the terminal box for detector (Note 1)
Water free treatment
Oil free treatment

Note)

Optional selections-

1. Must be selected for Tag no. requirement

24

Azbil Corporation

No. SS2-MGG200-0100

Submersible detector with FM NI approval
Wafer type (25 to 200 mm (1 to 8 inches)) Polyurethane rubber lining
Model MGG19D - I II III IV V VI VII VIII IX - X XI - Y / Options (some options can be selected per each model)
Basic model no.

Selections

MGG19D
I

II
III

IV

V

VI
VII
VIII

(1 inch)
25 mm
025(1½ inches)
40 mm
040
(2 inches)
50 mm
050
(2½ inches)
65 mm
065
(3 inches)
80 mm
080
(4 inches)
100 mm100(5 inches)
125 mm125(6 inches)
150 mm150(8 inches)
200 mm200
LiningQ
Polyurethane rubber
Wafer JIS 10K
Piping connection
11
Wafer JIS 20K
12
Wafer JIS 30K
13
Wafer ANSI 150
21
Wafer ANSI 300
22
Wafer JIS G3443-2 F12 (line size 80 mm or larger)
31
Wafer DIN PN10
41
Wafer DIN PN16
42
Wafer DIN PN25
43
Wafer DIN PN40
44
Wafer JPI 150
61
Wafer JPI 300
62
Electrode
SUS316L
L
TitaniumK
Tungsten carbide (only for size 10 mm)
W
Other
_
Grounding ring
SUS316
S
TitaniumK
Other_
Electrical connection Remote
G1/2 internal thread / with brass (Ni-plated) watertight gland/ watertight gland
type
G1/2 internal thread / with SUS304 watertight glandFace-to-faceStandard
dimensions
Upstream side (horizontal / vertical piping mounting)Installation / wiring Remote
direction
type
Downstream side (horizontal / vertical piping mounting)
Horizontal piping mounting / left side viewed from upstream
Horizontal piping mounting / right side viewed from upstream
Calibration
Standard+/- 0.35 % of rate calibration (Size 40 to 200 mm (1 1/2 to 8 inches))
Other

X
XI

FinishBolt / nut

Azbil Corporation version (must be selected)
Traceability certificate for detector
Material certificate (only for electrodes and ground rings)

Note)

selections-

Line size

Options

IX

Optional

-

3
8
A
A
B
C
D
A
U
_
X

Standard
None
Carbon steel
SUS304X
1
2

With gasket for plastic piping
Attachment of the TAG number to the terminal box for detector (Note 1)

Y
B
C
J
K

Water free treatment

E

Oil free treatment

F

1. Must be selected for Tag no. requirement25

No. SS2-MGG200-0100

Azbil Corporation

DIMENSIONS(Unit : mm (inch))

Union joint (size 2.5 to 15 mm (0.1 to 1/2 inch))

Hose joint (size 2.5 to 15 mm (0.1 to 1/2 inch))26

Azbil Corporation

No. SS2-MGG200-0100

IDF / Tri clamp (size 2.5 to 15 mm (0.1 to 1/2 inch))(Unit : mm (inch))

Note)
			

1. An integral detector includes an integral converter instead of a terminal box.
2. Clamp size: 1S

27

No. SS2-MGG200-0100

Azbil Corporation

Wafer type (size 2.5 to 15 mm (0.1 to 1/2 inch))

(Unit : mm (inch))

Wafer type (size 25 to 200 mm (1 to 8 inches))

Size
Face to facedimension

L
H1

HeightH2
H3
H4

Housingwidth
Housingouterdiameter
Weight
Note)

W
φD

mm

2.5

5

10

15

25

40

50

65

80

100

125

(inches)mm(inches)mm(inches)mm(inches)mm(inches)mm(inches)mm(inches)
mm

(0.1)
56
(2.20)
71(2.80)
72(2.83)
190(7.48)
96(3.78)
98(3.86)
49.5

(0.2)
56
(2.20)
71(2.80)
72(2.83)
190(7.48)
96(3.78)
98(3.86)
49.5

(3/8)
56
(2.20)
71(2.80)
72(2.83)
190(7.48)
96(3.78)
98(3.86)
49.5

(1/2)
56
(2.20)
71(2.80)
72(2.83)
190(7.48)
96(3.78)
98(3.86)
49.5(1)56(2.20)
77(3.03)
34(1.34)
158(6.22)
102(4.02)
106(4.17)
68(1½)
80(3.15)
84(3.31)
43.5(1.71)
175(6.89)
109(4.29)
125(4.92)
87(2)
86(3.39)
93(3.66)
52(2.05)
192(7.56)
118(4.65)
135(5.31)
104(2½)
96(3.78)
100(3.94)
62(2.44)
209(8.23)
125(4.92)
148(5.83)
124(3)
106(4.17)
108(4.25)
67(2.64)
222(8.74)
133(5.24)
164(6.46)
134(4)120(4.72)
121(4.76)
79.5(3.13)
247(9.72)
146(5.75)
189(7.44)
159

(5)
(6)
(8)
140
160
200
(5.51) (6.30) (7.87)
133
160
185
(5.24) (6.30) (7.28)
95
110
135
(3.74) (4.33) (5.31)
275
317
367
(10.83) (12.48) (14.45)
158
185
210
(6.22) (7.28) (8.27)
214
240
290
(8.43) (9.45) (11.42)
190
220
270

(inches) (1.95)(1.95)(1.95)(1.95)(2.68)(3.43)(4.09)(4.88)(5.28)(6.26)(7.48)(8.66) (10.63)

2.6
(5.7)

2.6
(5.7)

2.3
(5.1)

2.6
(5.7)

2.8
(6.2)

3.4
(7.5)

4.5
(9.9)

5.2
(11.5)

6.7(14.8)

10.0(22.0)

13.6(30.0)

kg
(lb)

2.6
(5.7)

1. An integral detector includes an integral converter instead of a terminal box.

28

150

200

22.0
(48.5)

Azbil Corporation

No. SS2-MGG200-0100

Flange type (size 2.5 to 15 mm (0.1 to 1/2 inch))

(Unit : mm (inch))Flange type (size 25 to 150 mm (1 to 6 inch))

Size
Face to facedimension

HeightWeightNote)

mm(inches)mm
L(inches)mm
H1(inches)mm
H2(inches)mm
H3(inches)mm
H4(inches)
kg
(lb)

2.5

5

10

15

25

40

50

65

80

100

(0.1)
160
(6.30)
71(2.80)
72(2.83)
190(7.48)
96(3.78)
5.0(11.0)(0.2)
160(6.30)
71(2.80)
72(2.83)
190(7.48)
96(3.78)
5.0(11.0)(3/8)
160(6.30)
71(2.80)
72(2.83)
190(7.48)
96(3.78)
5.0(11.0)(1/2)
200(7.87)
71(2.80)
72(2.83)
190(7.48)
96(3.78)
5.0(11.0)(1)200(7.87)
77(3.03)
63(2.48)
187(7.36)
102(4.02)
7.4(16.3)(1½)
200(7.87)
84(3.31)
70(2.76)
201(7.91)
109(4.29)
6.5(14.3)(2)200(7.87)
93(3.66)
78(3.05)
218(8.56)
118(4.65)
10.1(22.3)(2½)
200(7.87)
100(3.94)
88(3.44)
235(9.23)
125(4.92)
12.1(26.7)(3)200(7.87)
108(4.25)
93(3.64)
248(9.74)
133(5.24)
12.6(27.8)

(4)
(5)
(6)
250
250
300
(9.84) (9.84) (11.81)
121
133
160
(4.74) (5.24) (6.30)
105
125
140
(4.13) (4.92) (5.51)
273
305
347
(10.73) (12.01) (13.66)
146
158
185
(5.73) (6.22) (7.28)
18.4
26.0
30.6
(40.6) (57.3) (67.5)

1. This table is for remote detectors.

			

2. An integral detector includes an integral converter instead of a terminal box.

			3. The table indicates dimensions for ANSI 150 flange.

29

125

150

No. SS2-MGG200-0100

Azbil Corporation

Flange type (size 200 to 600 mm (8 to 24 inches))

Size
Face to facedimension

L
H1
H2

HeightH3
H4

Weight (kg)
Note)

(Unit : mm (inch))

mm

200

250

300

350

400

450

500

600

(inches)mm(inches)mm(inches)mm(inches)mm(inches)mm(inches)
kg
(lb)

(8)

(10)

(12)

(14)

(16)

(18)

(20)

(24)

350
(13.78)185(7.28)
196(7.72)
428(16.85)210(8.27)
48.0(105.8)450(17.72)235(9.25)
221(8.70)
503(19.80)260(10.24)60.0(132.3)500(19.69)258(10.16)250(9.84)
555(21.85)283(11.14)73.0(160.9)550(21.65)282(11.10)273(10.75)602(23.70)307(12.09)96.0(211.6)600(23.62)310(12.20)321(12.64)678(26.69)335(13.19)128.0(282.2)600(23.62)339(13.35)353(13.90)739(29.09)364(14.33)168.0(370.4)600(23.62)366(14.41)383(15.08)796(31.34)391(15.39)202.0(445.3)650(25.59)415(16.34)446(17.56)908(35.75)440(17.32)272.0(599.7)

1. This table is for remote detectors.

			

2. An integral detector includes an integral converter instead of a terminal box.

			3. The table indicates dimensions for ANSI 150 flange.

30

Azbil Corporation

No. SS2-MGG200-0100

Flange type (size 700 to 900 mm (28 to 36 inches))

(Unit : mm (inch))Flange type (size 1000, 1100 mm (40, 44 inches))

Size
Face to face dimension

L
H
H1

HeightH2
H3

Feet length

W

Feet widthA

Feet halls *n-φd

Weight
Note)
			

mm

700

800

900

1000

1100

(inches)mm(inches)mm(inches)mm(inches)mm(inches)mm(inches)mm(inches)mm(inches)mm(inches)
kg
(lb)

(28)
1100
(43.31)967(38.07)491(19.33)429(16.89)454(17.87)1049(41.30)600(23.62)4-φ33(1.30)
394(15.51)(32)
1200(47.24)1081(42.56)554(21.81)480(18.90)505(19.88)1147(45.16)600(23.62)4-φ33(1.30)
476.0(18.74)(36)
1300(51.18)1185(46.65)608(23.94)530(20.87)555(21.85)1245(49.02)600(23.62)4-φ33(1.30)
566(22.28)(40)
1500(59.06)1278(50.31)650(25.59)581(22.87)606(23.86)980(38.58)800(31.50)4-φ33(1.30)
823(32.40)(44)
1500(59.06)1399(55.08)720(28.35)632(24.88)657(25.87)1000(39.37)800(31.50)4-φ33(1.30)
930(36.61)

The table indicates dimensions for ANSI 150 flange.
*: n = number, d = diameter31

No. SS2-MGG200-0100

Azbil Corporation

MagneW™ is a trademark of Azbil Corporation in Japan and/or other countries.
Please read “Terms and Conditions” from the following URL
before ordering and use.

https://www.azbil.com/products/factory/order.html
Specifications are subject to change without notice.

1-12-2 Kawana, Fujisawa
Kanagawa 251-8522 Japan
https://www.azbil.com/(16)

1st edition: Feb. 1997
18th edition: May 2023

32

No part of this publication may be reproduced or duplicatedwithout the prior written permission of Azbil Corporation.