Vibration Analysis System for Elevator

MODEL-6601
Compliant with ISO18738-1

This vibration analysis system complies withthe elevator ride quality standards specified
by I SO. D at a are c olle c ted by t h eequipment placed on an elevator andanalyzed by a PC.

z Resolution: 0.005 m/s2 (0.5 gal)z Possible to issue a calibration certificate requiredby ISOz Appended with easy-to-use software for analyzing
elevator ride quality (in Japanese and English)
z 3.5" color LCD installed; waveform possible to bechecked on the main body

The easy-to-see color LCD enables
reduction of erroneous measurement.
As measured waveforms are displayed on the color LCD in real time, checking
whether measurement and recording are properly proceeding as expected isenabled during measurement.

z Highly accurate measurement enabled
Response axis:

3 axes (X, Y, Z)

Frequency range:

DC to 640 Hz

Max. acceleration: 5G (49.03 m/s2) = 4,903 gal
Resolution:

0.005 m/s2 (0.5 gal)(value required by ISO18738-1)

Max. recording length: 10 minutes
Sampling frequency: 250, 500, 1,000, and 2,000 Hz

z Easy to measure
The equipment is ready for conducting measurement only by placing it on an elevator and triggering
the measurement. Measured data are stored in theUSB memory and enabled with detailed analyses
on a PC.

Measurement style selectablein response to a particular situation
Measurement style (1)Measurement style (2)Measurement while monitoring waveforms
in real time is enabled.

Measurement is enabled with the detector installedinside the main body and the casing lid remaining
closed.
The equipment is operated by using the remote unit.

z Various analyses required by
ISO18738-1 can be easily conducted byusing the dedicated software.

Analysisz The equipment is useful for identifying causes of abnormal
vibration, as frequency analysis (FFT) is also enabled.

z The measurement reportrequired by ISO18738-1
can be easily prepared by
one click.

Specifications of the main body
Display

3.5" color LCD

Detector

3-axis acceleration detector; DC to 640 Hz (-3 dB)

AD converter

16 bits

Sampling frequency

Settable at 250, 500, 1,000, or 2,000 Hz.

Frequency range

Settable at DC to 80 Hz, DC to 160 Hz,
DC to 320 Hz, or DC to 640 Hz.

Max. acceleration

5G (49.03 m/s2) = 4,903 gal

Output±2V/full scale; BNC connector for each X, Y, and Z

Resolution

0.005 m/s2 (0.5 gal)Max. recording length

10 minutes

Recording means

Recorded in the USB memory appended.

Power supply

12V lithium-ion battery (continuous use of approx. 10 hrs)

Charging time

Approx. 10 hrs

Appearance of themain body and mass

300 (W) u 275 (D) u 110 (H) mm, 3 kg

Software specifications
Responding OS

Windows XP, Windows Vista, and Windows 7

Responding languages

Japanese and English

Report preparation

Responds to the requirement of “6.5 Reporting of results” inISO18738 -1.

FFT analysis function

Window function: Rectangular, Hamming, and HanningCan be analyzed with raw data, and results of RMS,Average, and Filter for each X, Y, and Z.

Computing function

RMS and Average
Combination of XY, YZ, XZ, or XYZ
Additions of HPF and LPF (arbitrary cutoff)

Display

Provided with screen zooming function.

Max. value detection

Equipped with automatic detection function.

Output

CSV file output

Configuration
Standard configuration

OptionVibration analysis system forelevator MODEL-6601

1 unit3-axis acceleration detector

1 pc

Calibration and traceability certificates 1 set

Detector extension cable

1 pc

USB memory

1 pc

Remote unit (with cable)

1 pc

Charger,

1 pc

Analyzing software (CD-ROM)

1 set

Inspection Results

1 set* Specifications and designs presented in the product catalog are subject tochange without notice for product improvement purposes.

http://www.showasokki.co.jp/

SHOWA SOKKI CORPORATION

Agent:

Head:1-5-9,Kandaizumi-cho,Chiyoda-ku,Tokyo,101-0024
HeadOffice
Office: 1-5-9,Kandaizumi-cho, Chiyoda-ku, Tokyo,
Tel.03-3866-3210(main)Fax.03-3866-3060
101-0024
JAPAN.
Osaka Branch Phone::Hiranomachi
Grand Building
6F+81-3-3866-3060
EAST,+81-3-3866-3210
Fax:
1-7-14,Hiranomachi,Chuo-ku,Osaka, 541-0046
Factory:
1547-1,
Takao-cho, Hachioji, Tokyo,
Tel.06-6205-5070(main)Fax.06-6125-5111
193-0844
JAPAN.
Factory:3-16-2,Sennin-cho,Hachioji-shi,tokyo,193-0835
Phone:+81-42-664-3232 Fax: +81-42-664-3276
Tel.042-664-3232(main)Fax.042-664-3276

E-mail: eigyo@showasokki.co.jp

13100.5K（RT）