SANTEC OPTICAL COMPONENTS

Multichannel Optical Power Monitor

MPA-100

The MPA-100 is a highly integrated rack model of
pigtailed photo-detector array. Multi-channels opticalpower from MPO or MTP connectors are detected byphoto-detector array. Photo-detector of Si type for
850nm range and InGaAs type for 1260~1610nm areavailable. The detected photo-current are converted tovoltage output using a 16-bit DAQ interface.
The dimension is (W)482.6x(D)430x(H)44 mm. The
highly integrated MPA-100 is available up to 80 channels.
The MPA-100 is useful for many applications, such as
testing and monitoring of transceivers for data
communications in data centers, broadband / CATV, andtelecommunications. Tap-PD model (fibers input&output,a partial power detection) is also available upon customer’s
request.

Features

Applications

Multi-channels photo-detector array up to 80 channels
Power monitoring and testing at 850 nm SR4, SR10
Wide-operating wavelength range: 1260 ~ 1610nm
Wide-optical input range: -50 ~ -10 dBm
Excellent linearity: +/- 0.6dB
MPO adapter / High-density MPO, MTP connectors
16-bit DAQ voltage output by a USB interface
1U Size (44mm) slim design
Custom model is available upon request

Optical power testing of transceivers for IEEE
802.3ba based SR4 ,SR10, LR4, ER4
Optical power monitoring for data communications,broadband / CATV networks, and telecommunication
networks
Optical power monitoring in data center equipment
Optical power monitoring of Interconnections ofactive optical cables.

Measurements
Example of input power and output voltage for 80
channels.

SANTEC OPTICAL COMPONENTS

Specifications
Item

Total optical input ports
Wavelength range

Min.
SR TypeIR TypeOptical input power range

Absolute max optical input power

Max.
80

32860

840

1260

1610

-+10-50-

Optical noise limit

Unitschannels
nm+3dBm-55dBmdBm

Optical ﬁber

MMF, SMF-

Optical adapter

MPO adapter-16bit

MPO, MTP 2)

Optical connector
DAQ

Data sampling rate

PD response bandwidth 1)

10Lineality

Digital Input & Output
Wavelength
calibration data-0.6

SR Type

Operating temperature

USB-YES-5

WeightWarranty

0.6

4.4

kHzdBm-30+35

15

Storage temperature-Samples/sNO

IR TypePower consumption

9,375+70

Notes

Options: 32, 48, 64, 72, 80
SR: Silicon photodiodeIR: InGaAs photodiodeper channel

Typ. -63 dBm@1550nm

MMF: Core dia. 50 mm, Clad dia. 125mm
SMF: ITU-T G.652

Gender: Pins(male), Polish: Angled polish (8°)
NI USB 6255 OEM
Typ. 30+/-3 kHz

Typ. +/-0.1 dB, optical input power range-Wavelength, optical power, output voltage℃

RH: 85%Max, No condensation

W℃kg

Typ. 20W, DC12V

RH: 85%Max, No condensation

1Year
USB、Input / Output Trigger : BNC connector x 2 portsExternal terminal

AC adapter, power cord (AC100/240V-DC12V), USB cable (2m), operation manual, certiﬁcate of test, control software

Accessories

1) The bandwidth is deﬁned as 3dB down (Linear scale: 0.7071) of optical frequency response. The response slope is designed in 20dB/dec.
2) Other adapters such as SC type and LC type are also available.

Dimensions

USBPFI 0PFI 1

44430

DC 12V430

Unit [mm]482.6

www.santec.com/en/2024© Santec AOC corporation santec reserves the right to make changes in equipment design, components or specifications without notice.

Santec Japan Corporation

5823 Ohkusa-Nenjozaka, Komaki, Aichi 485-0802, Japan Tel. +81-568-79-3536 Fax +81-568-79-1718

Santec USA Corporation

400 Kelby Street Suite 1501, Fort Lee, NJ 07024, USA Toll Free +1-800-726-8321(santec-1) Tel. +1-201-488-5505 Fax +1-201-488-7702

Santec Europe Ltd.

99 Park Drive Milton Park, Abingdon Oxfordshire, OX14 4RY, U.K. Tel. +44-20-3176-1550

Santec (Shanghai) Co., Ltd.

21F Room H, Hua Du Bldg., No.838 Zhangyang Road, Pudong District, Shanghai 200122 China Tel: +86-21-58361261
MPA-100-C-E / Ver.1.7 CODE-202409-HN-HN-CPY