GENERAL CATALOG
Vol.18.14

Manufacturer

AIKOH ENGINEERING CO.,LTD.

URL http://www.aikoh.co.jp/
TOKYO14-1, 5-Chome, ND bldg.,7F, Ueno, Taito-ku, Tokyo
TEL (03)5807-6434 FAX (03)3834-2098
NAGOYA 210, 2-Chome, Sakuta, Nagakute city, Aichi
TEL (0561)64-2331 FAX (0561)64-2332
OSAKA
15-7, 2-Chome, Hishie, Higashi-Osaka city, Osaka No.4420-ISO 9001
No.E2203-ISO 14001
TEL (072)966-9011 FAX (072)966-9017

Overseas sales dept.
Osaka oﬃce
15-7, 2-Chome, Hishie, Higashi-Osaka city, Osaka
TEL (072)966-9011 FAX (072)966-9017※ALL SPECIFICATIONS SUBJECT TO CHANGE PRIOR NOTICE.

160502000

Table of Contents

Pursuing Higher Reliability and Uniformity
from Customers' Standpoint

Knob evaluation test

On-Board Unit

Feeling test of powered windows

Various digital force gages

Feeling evaluation test of shift levers

Portable torque gages
Handy series force
measuring instruments

No matter how the times change, wish for safety and security remains

Tape adhesion test

Peeling adhesion test

Peel strength test of embossed tape

Bending test by setting angle

located in various places in the world and manufacturing activities diversified
and fragmented, Aikoh Engineering, since its foundation, has committed itself
to "quality" through measurement of load for more than 40 years. We haveoffered various services including physical property-to-environment composite
tests and physical property-to-electrical characteristics composite tests as wellas individual tests of tension, compression and rotation.

Shearing jig

Shearing jig〜10mm/min

P.6

Test stand (Vertical & Horizontal)

P.7, P.8

Measuring ampliﬁer

MODEL-1016C

P.9, P.10

Small size desk top
force tester

MODEL-1305VC

P.11

MODEL-1605VC
MODEL-1605VCL

P.12

FTN1-13A

P.13

Precision force tester

QFP

Shearing direction
Epoxy adhesive

Desk top typeforce tester

5〜30mm/min
Substrate

Die shearing test

Hooking45°

Chip parts

Solder shearing test of chip parts

Substrate

45°pulling test of QFP lead solder joints

Solenoid testof the reliability of products of our dear customers through our experience andknow-how accumulated over a long period of time.

Large size force testers

LCD surface

We would like to ask for your continued business and support in the future.

Feeling test of tact switches

Bending test by setting displacement

Insertion & withdrawal test of connectors

Insertion & withdrawal test of connectors

Switch feeling testers
Horizontal type force
testerSmall size separation
evaluation tester
Embossed carrier tapeseparation tester

Executive President

Kazuya Yoshioka

Precision force testers
Large size precision
force testers
Large size desk top
type force testers

All of the members at Aikoh are determined to be of assistance to enhancement

Durability test of mobile devices

Feeling measurement of electric parts

Materials tensile test

Sheet urethane rubber test

● Screen display type force measuring● Handy force measuring● Torque - angle measuring instruments● Various load cells● Automatic force - displacement● Various assembling jigs

instruments

measuring instruments● Various testers for durability

evaluation

Spring test

Tablet rupture test

Cut strength test of ampules

Peeling test

instruments● Spring testers

Buckling test

Hardness and stress relaxation measurementof jelly,butter,bread,etc.

Opening force test

Shearing test of boiled ﬁsh paste, hams, etc.

Unpacking force test

Bending test

Immersion elasticity test of fruits, etc,

Insertion & withdrawal test of key cylinder

P.14
P.15, P.16

MODEL-1320 Series

P.17

MODEL-1431VC Series

P.18

GT-FL500
MODEL-1616W
MODEL-2152VCE
MODEL-2252RDH

P.19
P.20
P.21

MODEL-2165P

P.21

Adhesion tester

FTN4-15A

P.22

Touch panel typespring testers

SHR Ⅲ Series

P.23, P.24

MODEL-5125VC/VCW
MODEL-5125VCT

P.25

MODEL-5401VC-50/200

P.26

MODEL-5127VC/500 〜 5000

P.26

MODEL-1605VC/NF
MODEL-1308U/NF

P.27

Custom order products

Comprehensive syringe needle force
measuring tester
Power cable bending durability tester
Switch & silicone rubber durability tester

P.28

Grips & Jigs

ChucksGrips
Vises
X-Y tables

P.29, P.30

Indicators

MODEL-0218B
MODEL-0215T

P.31

Load cells

Various load cells

P.32

Standard softwares

FS-700, FSN-500, RZ-3000, FL-2005

P.33, P.34

Large size torqueangle testers
JIS-rated lead freesolder testers● Design of measurement

software

MODEL-1310VC/VCW
MODEL-1311VC/VCW
MODEL-1840VC Series
MODEL-1840VCT Series

MODEL-1308UH

Torque angle testers

Product Lineup

Switch feeling instruments
Bench type simple tester MODEL-1309RZ

unchanged and concerns about quality keep growing.
In the industrial world where Internet has been spreading, production bases

P.3 〜 P.5

RZ-T Series Portable Torque Gages

RZ Series Digital Force Gages

RZ-T Series Standard Specifications
Model
Max. force

RZ-T-20

Min. display unit

RZ-T-100

2000mN・m10N・m(20kgf・cm)(100kgf・cm)

1mN・m

0.001N・m(0.01kgf・cm)(0.01kgf・cm)

mN・m, kgf・cm
N・m, kgf・cm
Unit of measurement±1%
Accuracy
1 ms, 5 ms, 16 ms, 50 ms, 125 ms and 250 ms selectable
Sampling cycle
1, 2, 5, 10, 20 times/sec. selectable
Display update interval

The RZ Series digital force gages are newly developed user-friendly gages that have significantly improved theoperability of conventional push-pull gages while keeping their functions.
They come with such convenient functions as data output by use of a newly adopted USB and charging and theyalso feature quick and easy operations such as changing sampling rates, changing over between three units, printeroutput and display reversing.
Their shape and buttons conventional concept.

RZ Series Model List
Model
Rated capacity(R.C.)

RZ-1

RZ-2

10N

20N

RZ-5

RZ-10

50N

100N

RZ-20

200N

RZ-50

The portable torque gages RZ-T Series are especiallydesigned for static measurements such as screw
retightening and returning torque, breakage torque due totwisting, etc. With a large and easy-to-see display,
simultaneous reading of clockwise and counterclockwise
torque, NO-GO-NO judgment function and residual battery
capacity display function, the RZ-T Series are state-of-the-art
multi-function torque gages. The bit or jig at the top ischanged according to the measurement requirements.
Measured data may be stored in the memory or output to aconnected printer. Two measurement units are available:
mN・m and Kgf・cm.

1000N（1Kgf）（2Kgf）（5Kgf）（10Kgf）（20Kgf）（50Kgf）（100Kgf）

0.001-10.000N

0.01-20.00N

0.01-50.00N

0.01-100.00N

0.1-200.0N

0.1-500.0N

0.1-1000.0N

Indication range （0.1gf-1.0000Kgf） （1gf-2.000Kgf） （1gf-5.000Kgf） （1gf-10.000Kgf） （10gf-20.00Kgf） （10gf-50.00Kgf） （10gf-100.00Kgf）
Min. indication 0.001N（0.1gf）
unit

0.01N（1gf）

0.1N（10gf ）I/O cable
RZ-OP-1
I/O cable
RZ-OP-2
MITUTOYO-made DP-1VR printer cable
RZ-OP-3
SANEI ELECTRIC-made line thermal printer cable RZ-OP-4

USB output
Analog voltage ±2V/F.S.
Printer output
Motorized stand control
External contact hold30 hours after full charge

Continuous use timePower supply

Nickel-metal hydride battery
AC adapter Model 780

MassStandard accessories
Options

Approx. 375g
Display: RZ-S, Torque meter,
Standard attachment,
AC adapter MODEL-780
Software RZ-3000,
I/O cable RZ-OP-1,I/O cable RZ-OP-2,
Mitutoyo DP-1VR printer cable RZ-OP-3,
Sanei Electric line thermal printer cable RZ-OP-4

A handy type digital force gage with a measuring part anddisplay part separated. The separate construction poses
little restriction on measuring places to select.

RZ Series Standard Accessories

RZ Series Optional Cables

LCD, signed, 5 digits

OutputRZ-S-8000 Series Separate Type Digital Force Gages

RZ-100500N

Display

Please purchase the optional cable
RZ-OP-1 when you need only thestand control function.
Please purchase the optional cable
RZ-OP-2 when you need the analog
output and comparator output in
addition to the stand control function.

RZ-S-8000 Series Standard Accessories

RZ Series Standard Specifications
Unit of measurement
Accuracy
Allowable overload
Measurement system

N or (gf) Kgf, N, lb selectable

RZ-S-8000 Series Standard Specifications

Within ±0.2% of rated capacity
120% of rated capacity (Overload alarm at about 110%)

Model
Rated capacity
Min. display unit

Track mode/Peak hold mode/
Compression-tension peak hold mode selectable

Indication update cycle
Sampling cycle
Working temperature range
Guaranteed temperature range
Continuous working hours
Power supply

1, 2, 5, 10, 20 times/sec. selectable
1ms, 5ms, 16ms, 50ms, 125ms, 250ms selectable
0 to +40℃+5 to +40℃
30 hours after full charge
Dedicated nickel-metal hydride AAA battery x3
Dedicated AC adaptor: MODEL-780 (DC5V/1000mA)
Dedicated USB cable: MODEL-RZ-USB

External dimensions
Mass

MODEL-2257 (Horizontal type)W68xD40xH232mm
Approx. 375gs

MODEL-2256 (Vertical type)

RZ-S-8001

RZ-S-8002

RZ-S-8005

RZ-S-8010

RZ-S-8020

RZ-S-8050

RZ-S-8100

10N（1Kgf）

20N（2Kgf）

50N（5Kgf）

100N（10Kgf）

200N（20Kgf）

500N（50Kgf）

1000N（1Kgf）

0.001N（0.1gf）

0.1N（10gf）

0.01N（1gf）

Unit of measurement
Accuracy
Sampling cycle

1 ms, 5 ms, 16 ms, 50 ms, 125 ms and 250 ms selectable

Display update interval
OutputUSB output, Analog voltage ±2V/F.S., Printer output, Motorized stand control, External connection holdN or (gf) Kgf, N, lb selectable±0.2%F.S.
1, 2, 5, 10, 20 times/sec. selectable

Usable timePower supply

Nickel-metal hydride battery, AC adapter MODEL-780 (AC100-240V)

Mass
Standard accessories

Display RZ-S, Load cell MODEL-S8000 series with cell cable (3 m), Standard attachment, AC adapter MODEL-780(USB cable : includes MODEL-RZ-USB)

Options

30 hours after full charge
Approx. 375g （Main unit only）
Software RZ-3000, I/O cable RZ-OP-1, I/O cable RZ-OP-2, Mitutoyo DP-1VR printer cable RZ-OP-3
Sanei Electric line thermal printer cable RZ-OP-4, Printer BL2-58SNWJC, jig/fixture, test standRZ-OP-1 : For Stand control function only
RZ-OP-2 : For Stand control function, Analog output and Comparator output

SX Series Digital Force Gages

Bench Type Simple Tester

SX Series Standard Specifications
Measurement unit
Measurement item
Accuracy
Sampling frequency
Display update interval
Display
FunctionsStandard
accessories

This is a force gauge with the functions required for a
handy-type test equipmet.Operation is easy for
simultaneous display of the compression and tensile force
values.External inputs allow easy printout. (A printer is
optional.)
The AC adaptor exclusive for the SX is applicable to100-240V with simple plug replacement allows uses in awide area.

SX Series Model List
Model

SX-2

Max. force（2Kgf）

Min. display unit

20N

SX-5

50N

（5Kgf）

0.01N/0.001Kgf/1gf

SX-20

200N

（20Kgf）

SX-50

500N

（50Kgf）

Options
Applicable temperature range
WeightPower supply

N, Kgf (gf), Lbf selectable
Track/Peak/+Peak/-Peak±0.2% F.S.

MODEL-1309RZ
Capacity: 1000N (100 kgf)
Force - Displacement

/'

1

||

20, 62, 200 & 800 times/sec. selectable

Materials tensile test

1,2,5,10 & 20 times/sec. selectable
LCD, Signed 5-digits, 40 x 40mm

6 attachments, exclusive AC adaptor (100 to 240VAC)
and exclusive carrying case
Thermal printer BL2-58 & cables

RX-FL Series Optional Cables
Analog output cable
RS232C interface cable
MITUTOYO printer cable
External control cable

RX-OP-1
RX-OP-2
RX-OP-3
RX-OP-5

Bending test

Standard Specifications

0 to 50℃

Max. force
Force display
Test speed
Speed changing

Approx. 360g
Three exclusive AAA nickel hydrogen cells,
AC adaptor MODEL-770 (5VDC, 1200mA)

SX Series Standard Accessories

1000N (100Kgf)
Depending on RZ Series
10 to 300 mm/min.
LO: 10mm/min.
MD: 100mm/min.
HI: 200mm/min. Intended
setting is available.
VR: 10 to 300mm/min.

400mm
StrokeDisplacement display (Resolution) 0.1mm
Displacement display (Max. display) ±400mm
295 x 175mm
Table size
Approx. 35Kg
Weight
W300 x H850 x D340mm
Size
100 to 120VAC/200 to 240VAC
Power supply

SX Series Optional Cables
Over load cable
Printer & External input cables

SX-OP-1
SX-OP-2

0.1N/0.01Kgf

Standard Configuration
Tester:MODEL-1309RZ
Application
for MODEL-1309RZ:
Standardsoftware
Configuration
RZ-3009
Cable for connecting RZ Series: RZ-OP-8
USB cable

The RX-FL Series measures the peak values, bottom
values, click ratios, click values, etc. of silicone rubber anddome switches in a single sequence of operation.It has alarge screen and stores measured data in memory, which
may be printed out or output to a PC after measurement.

RX-FL Series Standard Specifications※The manual stand MODEL-1338 is optional.

Shearing test of boiled
ﬁsh paste, hams, etc

A small stand for force-displacement measurement dedicatedto RZ Series. With a displacement resolution of 0.1mm, thisstand is suitable for long-stroke tensile/compression tests. A
force-displacement FS curve can be
drawn on a PC screen by the dedicatedsoftware for effective data management.

External printer input, external contact hold input,
external zero-reset input, overload output,
thermal printer BL2-58 printer output,
reset after printing (ON/OFF), automatic power off (ON/OFF),
automatic zero reset (ON/OFF),both compression peak and tensile peak display

Caution)The min.display unit is not accuracy.

MODEL-RX-FL-1・RX-FL-2

\L

II,

iii

£11

Model
RX-FL-1
RX-FL-2
Measurement range
10N (1Kgf)
20N (2Kgf)
Min. display unit
0.001N (0.1gf)
0.01N(1gf)
N, Kgf, Lbf selectable
Unit of Measurement
Peak value, Bottom value, Click value & Click ratio
Measurement item
±0.2%F.S.
Accuracy
5ms (100Hz)
Sampling frequency
6-digits LCD, 44 x 44mm
Display
16-bit, 100kHz
A/D converter
16-bit, single-chip CPU
CPU in use±2V/F.S.
Analog output
RS232C, 38400bps
Communication output
SANEI thermal printer, MITUTOYO DIGIMATIC output
Printer output
Lamp indication & output
GO-NG judgment
Stand connection output Overload, zero reset & external input display hold contact
Approx. 12hours
Continuous run time
0 to 40℃
Applicable temperature range
Approx. 450g
WeightNicd & AC adaptor MODEL-761 (100VAC) or MODEL-762 (200-240VAC)
Power supply※Digital force gage RZ Series is optional.

Dedicated application software RZ-3009

Major functions
Item
Data acquisition
Data file processing
Graph overlap display

Description
Real time force/displacement acquisition, graph drawing
Data save/batch save

Dedicated format/CSV format

Data load/batch load

Dedicated format

Number of registration

100

Graph cut sheet print

Cut sheet horizontal: 1graph
Cut sheet horizontal: 1graphPrinting

Graph continuous print

Cut sheet vertical: 2graphs
Cut sheet horizontal: 4graphs
Cut sheet vertical: 8graphs

Pick up registered data

※This software is compatible with Windows 7or 8 & Japanese or English OS.

Pick up registered data
Measurement list
display
Graph data list display
Stand control
USB serial conversion

Number of registration

100/measurement

Representative value detection Force max./min./average
All measurements processing

Force max./min./average

Time/displacement/force data values
Stand start/stop during test
I-O DATA: RSAQ6 (Recommended/option)

Desk-top type Testers

Wire Harness Withdrawal Force Tester
A long-stroke automatic test stand dedicated to RZ Series & SX Series. This is areasonable test stand that is suitable for tests of specimens of large size and long test
stroke. When used together with RZ Series or SX Series, such operations are possible
as overload monitoring and automatic stop and reversing when the load set in RZ Series
or SX Series is reached. For combining this stand with RZ Series or SX Series, please usean optionally available RZ-OP-1/RZ-OP-2 for RZ Series or SX-OP-1 for SX Series.

MODEL-1308U

The MODEL-2254 is a manual test stand for measuring thecaulking strength of wire harnesses and so forth. The work
mounting head is moved with a lever. Using the AIKOH RZ Series
digital gages provides various functions, including data printing andstorage in personal computers. The RZ Series digital gages are
optional. The MODEL-2254 is supplied with a set of gages jigs.

MODEL-2254

Standard Specifications

OptionGrip mounting plate
MODEL-OJ-P-90※Digital force gage is optional.

※Digital force gage is optional.

MODEL-2252R（Needed when using
the tensile test jig.）

Max. force
Test speed
Speed changing
StrokeDisplacement display
Distance between test center & column
Table size
Sample table size
Drive motor
Driving mechanism
Driving mechanism guide system
WeightSizePower supply

Standard Specifications
1000N (100 kgf)
Continuous
400 mmNone※Digital force gage is optional.

60 mm
W250 x D140 mm
W120 x D90 mm, center: M6
DC brushless motor
Trapezoidal screwLinear ball bush
Approx.20 kg

The MODEL-1349 is a lever type
Stand exclusively designed forcompression tests. The gage
mounting unit is moved downwith the lever.

The MODEL-1345 is a manual
test stand. The gage mounting
head is moved up and down by
rotating the handle.

Standard Specifications

Standard Specifications

Max. force
StrokeTable size
Max. span
WeightSize

Max. force
StrokeTable size
Max. span
WeightSize

43mm/140 degrees
180 x 100 mm
210 mmApprox.9kg
W200 x H435 x D235 mm※Digital force gage is optional.

500 N (50 kgf)
43mm (1.75 mm/rotation)
180 x 100 mm
210 mmApprox.9kg
W200 x H435 x D235 mm（Needed when using
the tensile test jig.）

W525 x H150 x D200 mm

Max. force
Moving distance
StrokeWeight (Vertical)(Horizontal)
Size

500 N (50 kgf)
3 mm/handle rotation
240 mmApprox.16kg
Approx.12kg
W200 x H500 x D150 mm (including handle)

Vertical & Horizontal Motorized StandA vertical and horizontal small automatic stand dedicated to RZ Series and SX Series. Thisis a reasonable test stand that is suitable for tests of specimens of small size and short test
stroke. When used together with RZ Series or SX Series, such operations are possible as
overload monitoring and automatic stop and reversing when the load set in RZ Series or
SX Series is reached. For combining this stand with RZ Series or SX Series, please use an
optionally available RZ-OP-1/RZ-OP-2 for RZ Series or SX-OP-1 for SX Series.

MODEL-2257

OptionGrip mounting plate
MODEL-OJ-P-90

500 N (50 kgf)

（Needed when using
the tensile test jig.）

10 to 80 mm/min.
In five steps or continuous
150 mmNone
Adjustable with 40 mm
39 to 79 mm
W100 x D100 mm
Trapezoidal screw
DC brushless motor
Approx.23 kg
W518 x H228 x D345 mm
100 to 240 VAC, single-phase, 0.5 A

Approx.15kg

Standard Specifications※Digital force gage is optional.

Standard Specifications※Digital force gage is optional.

0 to 10 mm

OptionGrip mounting plate
MODEL-OJ-P-90

The MODEL-2252R employs a motor-driven gage mount, which moves tothe right and left. Zstage may be adjusted vertically. The work mount table
has slotted holes for adjusting the jig position back and forth.
Max. force
Test speed
Speed changing
StrokeDisplacement display
Z stage stroke
Distance between test center & Z-axis plane
Table size
Driving mechanism
Drive motor
WeightSizePower supply

70mm

A vertical and horizontal manual handle type small standdedicated to RZ Series and SX Series. Capable of measurement
up to 500N. The gage mounting part can be moved by turning
the handle.

MODEL-2256

100 to 240 VAC, single-phase, 0.5 A

MODEL-1345

500 N (50 kgf)

Vertical & Horizontal Manual Stand

W255 x H810 x D300 mm

MODEL-1349

500 N (50 kgf)

Max. force
StrokeChuck width
WeightSize

5 to 100 mm/min., variable※Digital force gage is optional.

Standard Specifications（vertical type）
Max. force
Test speed
Speed changing
StrokeDisplacement display
Distance between test center & column
Table size
Drive motor
Driving mechanism
Driving mechanism guide system
WeightSizePower source

500 N (50 kgf)
10 to 200 mm/min.
Continuous
150 mmNone
49 mm
W200 x D120 mm
DC brushless motor
Trapezoidal screw
Sliding contact
Approx.20 kg
W215 x H380 x D235 mm
100 to 240 VAC, single-phase, 0.5 A

Waveform Display V Series Force Measuring Amplifier
The MODEL-1016C is a digital force measuring amplifier with an LCD
touch panel. Various measuring conditions are set by touching the LCD
panel. The MODEL-1016C has four-quadrant operations necessary forreciprocal measurement without the need of any CF cards, in principle.
It features remarkably improved graph processing speed and as highcommunication rate as 230.4 kbps. This amplifier stores all measured
data digitally and allows the user to reproduce data displayed on thescreen as many times as needed during measurement and after
measurement. It requires, however, an exclusive CF card for switch
feeling tests, separation tests and creep tests.

MODEL-1016C

Features

CF Card Types● V-103A
For switch feeling tests
Capable of setting the point - displacement test conditions,masking function and end stroke.
● V-108A
For separation tests● V-115A
Force holding creep tests
Force - displacement, and force - time

Hard copy

CF card● DOS Printer● Hard copies of the screen● Outputs text data of the waveform

graph

Analog output

RS232CPC

Stand connector

Control I/O connector

AC Outlet

Creep control output

Load cell input

X-Y recorder output

Analog direct output

Printer output

RS232C connector
for DOS/V

Specifications of MODEL-1016C● LCD touch panel● Capable of force - displacement measurement and torque - angle measurement.
● Offers four-quadrant operations necessary for reciprocal measurement.
● Features as high communication rate as 230.4 kbps.
● Able to calibrate the loads of a maximum of ten load cells.
● Able to read the force and displacement at any points with the cursor.
● Able to change the scales of the force and displacement graphs during measurement.
● Compact size for space saving● Save the data to the CF card. (.CSV )Connection with PC & Recorders

Linear gauge connector

Names

Load cell input

Load cell amplifier

Load cell application voltage: 12/6/3 Vdc (110 mA max., 350Ωx 3)
Voltage variation:
100 ppm/℃ max.
Input voltage range:
BV = 12V, 0.1 mV/V to 2 mV/V
BV = 6V, 0.2 mV/V to 4 mV/V
BV = 3V, 0.4 mV/V to 8 mV/V
Accuracy non-linearity:
0.01 % max.
Zero point movement:
1 μV/℃ RTI max.
Gain variation:
100 ppm/℃ max.
Low-pass filter frequency:
1, 3, 10, 30, 100, 300, 1k

A/D converter

Resolution:
Sampling frequency:

Length measurement counter
for rotary encoder & linear gauge
Proportional voltage output (Vp)
Voltage output for recorder(X.Y)
Control voltage output (Vref)
Control contact I/O

Load Cells & Force Display
Capacity

Display

Min. unit

5N (500g)±5000mN

1mN

Stand control output
Digital I/O

20N (2kg)±20.00N

0.01N

50N (5kg)±50.00N

0.01N

200N (20kg)±200.0N

0.1N

500N (50kg)±500.0N

0.1N

Operation panel
Source voltage

2KN (200kg)±2.000KN

1N

5KN (500kg)±5.000KN

1N

10KN ( 1ton)±10.00KN

0.01 KN

20KN (2ton)±20.00KN

0.01KN

50KN (5ton)±50.00KN

0.01 KN

100KN (10ton)± 100.0KN

0.1KN

Torque Meters & Torque Display
Capacity

Display

Min.unit

0.2N·m (2kgf・cm)

200.0mN・m

0.1mN・m

0.5N·m (5kgf・cm)

500.0mN・m

0.1mN・m

2N·m (20kgf・cm)

2.000N・m

0.001N・m

5N·m (50kgf・cm)

5.000N・m

0.001N・m

20N·m (200kgf・cm)

20.00N・m

0.01N・m

50N·m (500kgf・cm)

50.00N・m

0.01N・m

200N·m (20kgf・m)

200.0N・m

0.1N・m

500N·m (50kgf・m)

500.0N・m

0.1N・mPerformances & Specs.
Input channel: 1
Calibration points: 10

Display

Power consumption
Outside dimensions
Weight16 bits, sequential proportional type1000 times/sec. (1 ms)

Phase A/B up/down counter x 2 channels
Resolution: 24 bits
±10V, load resistance:
10kΩ or more
Output for monitoring load cell conditions±10V, load resistance:
10kΩ or more x 2 channels (X- & Y-axes)
D/A converter
Resolution: 12 bits
Updating frequency: 1000 times/sec. (1ms)±10V, load resistance:
D/A converter

10kΩ or more
Resolution: 12 bits

Inputs: 8
Outputs: 8 (relay contacts)
STOP, UP, DOWN & QUICK
Centronics-compatible parallel port for connecting a printerStart-stop synchronous serial port for connecting a PC
Baud rate: 4.8 k, 9.6 k, 19.2 k, 38.4 k, 115.2 k, & 230.4 k (bps)
STN type black-and-white LCD panel, 320 dots x 240 dots
Effective display area: 96 x 72 mm
10 X 6 touch panel
100 VAC ±10 % / 220 VAC ±10 %AC outlet (unswitched)
30 VA
W260 x H132.5 x D280 mm
Approx.5.2kg

CF card for Storing Important Data
V-103AFor switch feeling tests

V-108AFor separation tests

V-115AFor force holding creep tests

Small Size Desk Top Force Tester

Precision force testers

MODEL-1305VC

MODEL-1605VC / MODEL-1605VCL

Capacity: 200 N（20 kgf）
Force - Displacement

Capacity: 500N (50Kgf)
Precison force-Displacement
Touch panel load measurement

LCD surface

Tact switch feeling test

Tact switch feeling test

Bending test at preset displacement

Standard Specifications
Max. force
Test speed
Speed changing
StrokeDisplacement display
Detector
Display resolution
Display accuracy
Max. display value
Distance between test center & column
Table size
Drive motor
Driving mechanism
Driving mechanism guide system
WeightSizePower supply

200N(20kgf)
2 to 60mm/min.
In five steps or continuous
150mm
ProvidedLinear gauge
0.001mm0.05mm30.000mm
77mm
W330×D150mm
DC brushless motor
Trapezoidal screwLinear ball bush
Approx.32kg
W340×H580×D430mm
100 to 240 VAC,single-phase,0.5A

Measurement point
Setting

This is a small force tester designed for compression andtensile force tests up to 200N.It consists of a main unit
and MODEL-1016C amplifier and is capable of testingfeeling of such switches as tact switches and rubberswitches by using an optionally available CF card V-103A.

Peak Advance value
Bottom Advance value
Peak Return value
Bottom Return value
Force difference
Click rate
Contact ON
Contact OFF
Cursor position(CPF)ForceN・(CPT)Displacement mm(CLF)ForceN・(CLT)Displacement mm(RPF)ForceN・(RPT)Displacement mm(PLF)ForceN・(RLT)Displacement mm
CPF-CLF or CPF-RLF(CPF-CLF)/CPFx100 or (CPF-RLF)/CPFx100
ForceN・Displacement mm
ForceN・Displacement mm
ForceN・Displacement mm

When using CF card (V-103A)

MODEL-1605 Series are highly versatile force testers capable of high precision tests.These testers can perform
force-displacement tests under various conditions.They are equipped with such functions as detailed motion setting
required for various force tests and pickup data management.They are suitable for testing connector mating / unmating,metal characteristics and penetration.Tensile/compressive tests in a specific environment may also be
conducted.(Please contact us for a temperature range.)

Standard Specifications

On/Off Point Detector

For detecting On/Off points of silicon rubber switches* Jigs are not included. (Optional)

Standard Specifications
Input range
1, 10 & 100 kΩ (Selectable ranges)
Setting device 10-rotation helical potentiomeler with 500 even division scale
Setting accuracy ±0.5% in each resistance range (including non-linearity & hysteresis)
Output0 to 10V in each resistance range,load resistance:10kΩ
(1) Analog voltage(2) Monitor lamp

Red LED (Lit below set value.)(3) Open collector

Withstand voltage:35V max,suction current:50mA max.
ON voltage:1.5V max.

Responsedelay time
Temp.setting
Power supply
Size

ON: 0.5 mS or less - Until open collector output goes on after lowering below set value
OFF: 2mS or less - Until turning off open collector output after exceeding set value
0 to 40℃, no dew condensation
12 VDC (including 9 to 16.5 V ripples), 300 mA
140W×45.5H×140Dmm(including projections),approx.450g(not including AC adapter)

Model
Max. force
Test speed
Speed changing
StrokeDisplacement display
Detector
Display resolution
Display accuracy
Max. display value
Distance between test center & column
Table size
Driving mechanism
Drive motor
WeightSizePower supply

1605VC

1605VCL

500N (50Kgf)
0.5 to 600mm/min.

In fifteen steps or continuous
250mm

Tester with thermostatic oven image

Provided
Rotary encoder

Linear gauge

0.01mm0.001mm0.05mm±30.000mm (When using linear gauge)±250.00mm

100mm
W475 x D150mm M10x1.5
Ball screw
Servo motor
Approx. 50Kg
W480 x H578 x D375mm

W480 x H882 x D375mm

100 to 240VAC, single-phase, 5AStandard Configuration
MODEL-1605VC
Tester: MODEL-1605VC
Measuring amplifier: MODEL-1016C
Load cell: 1pc (Up to 500N)
MODEL-1605VCL
Tester: MODEL-1605VCL
Measuring amplifier: MODEL-1016C
Load cell: 1pc (Up to 500N)
Linear gauge: 1pc

Desk-top type Force Tester

Precison force testers

FTN 1-13A

MODEL-1310VC Series

Capacity: 500N/2kN (50Kgf/200Kgf)
Force-Displacement

Capacity: 2kN (200Kgf)
Force-Displacement
Materials tensile test

Insertion & withdrawal test of connectors

Bending test

lnserition & withdrawal test of connectors

Bending test

Materials tensile test

Withstanding load measurement of LCD

This is an integrated force tester with a built-in amplifier for the compression andtensile force tests up to 2kN (200Kgf).Two measurement speeds are available
according to the load cell capacity (300mm/min. for 500N, and 125mm/min. for2kN).A rotary encoder is provided for displacement detection.7-segment LEDs are in
use for data display.A software program is provided as the standard for easy tester
operarion and result output.Test conditions can easily be set by anybody.

Insertion & withdrawal test of connector

Standard Configuration
MODEL-1310VC
Tester: MODEL-1310VC
Measuring amplifier: MODEL-1016C
Load cell: 1pc (Up to 2kN)
MODEL-1310VCW (Wide range)
Tester: MODEL-1310VCW
Measuring amplifier: MODEL-1016C
Load cell: 1pc (Up to 2kN)
MODEL-1311VC
Tester: MODEL-1311VC
Measuring amplifier: MODEL-1016C
Load cell: 1pc (Up to 2kN)
MODEL-1311VCW (Wide range)
Tester: MODEL-1311VCW
Measuring amplifier: MODEL-1016C
Load cell: 1pc (Up to 2kN)

Features● The max. allowable test force is 2kN regardless of its compact size.
● Two measuring speeds 5 to 300 mm/min. (500N) and 5 to 125 mm/min. (2kN) areprovided.
● Compression, tension and breakdown tests are possible.
● The resolution is high at 0.1N for the force test and 0.01mm for the displacement.
● The power supply is 100VAC to 240VAC for worldwide uses.
● Simply operable personal computer software is provided as the standard.
● The load cell allows 3-ch calibration. (Optional feature)※Jigs are not included. (Optional)

Standard Specifications
Item
Max. force
Max. speed range
Moving speed range (inching & return)
Speed resolution
Resolution Force
5000display
2000display
1000display
Displacement
Accuracy
Force
Displacement
Driving block Motor type
Motor control system
Applicable load cell
Detector
Input/output Digital input/output
Analog input/output
Table size
Emergency stop
Outside dimensions
WeightPower supply

FTN1-13A
500N (50Kgf)

2kN (200Kgf)

5 to 300mm/min.

5 to 125 mm/min.

5 to 300mm/min.
0.1mm/min.
4-digit display
4-digit or 5-digit display
4-digit or 5-digit display
0.01mm3000 Series: ±0.2% F.S./UP Series: ±0.3% F.S.
±0.2mmPerformances
FTN1-13A
Item
Measurement start trigger Force/Displacement (LV setting available)
Force (Return/Stop, with a setting)Comparator
Displacement (Return/Stop, with a setting)Automatic zero
Manual zero return
Breakdown
Automatic return
Data detection

USB (For connecting to external PC)
Force (±10V/10bits)
Approx. 174 x 200mm
Installation on main body, driving block power off
W300 x H820 x D400mm
Approx. 40Kg
100 to 240VAC, single-phase, 3A

Breaking position detection (Detection sensitivity settable)
Comparator/hardware limit (with speed setting)
Measurement positive peak detection (Force/displacement)
Turn-back point (Force/displacement)

Pulse
Rotary encoder

Zero resetting by key operation

Measurement negative peak detection (Force/displacement)

Stepping motor
UP Series & 3000 Series (Up to 2kN)

Upon receiving measurement start trigger

Measurement start point (trigger position) (Force/displacement)

Automatic repetition of measurement 1 to 999999 times
Real-time data output Digital data output

Softwares
MODEL

FTN-3000

FTN-3001(Packed version)(Full spec version)

DOS/V compliant IBM/PC compatible
Computer
Operation system (OS) Windows 7 or 8

※Jigs are not included. (Optional)

These are all-round desktop type precision force testers for force and displacement measurements up to 2kN (200Kgf).
When combined with MODEL-1016C amplifier, they can measure two-point data of force-displacement andbreaking force-displacement.
A force-displacement graph is shown on an LCD panel in real time.
The two-column type is highly rigid, which makes these testers suitable for highly precise testing.
Standard Specifications
Model
Max. force
Test speed
Speed changing
StrokeDisplacement display
Detector
Display resolution
Display accuracy
Max. display value
Column length
Column interval
Table size
Driving mechanism
Drive motor
WeightSizePower supply

1310VC

1310VCW (Wide range)

1311VC

1311VCW (Wide range)

5 to 125mm/min.

0.2 to 250mm/min.

2kN (200Kgf)
5 to 125mm/min.

0.2 to 250mm/min.
In five steps or continuous
150mm
Provided

Rotary encoder

Linear gauge

0.01mm0.001mm

0.1mm

0.05mm±150.00mm±30.000mm (Using linear gauge)
548mm
250mm
φ200mm CenterM10 x 1.5
Ball screw

DC brushless motor

Servo motor

DC brushless motor

Servo motor

Approx. 41Kg
W356 x H900 x D314mm
100 to 240VAC, single-phase, 1A

100 to 240VAC, single-phase, 5A

100 to 240VAC, single-phase, 1A

100 to 240VAC, single-phase, 5A

Large Size Precision Force Testers

Large Size Precision Force Testers

MODEL-1840VCTMODEL-1840VC

Capacity: 2 to 50 kN (200 to 5000 kgf)
Force - Displacement

Capacity: 2 to 50 kN (200 to 5000 kgf)
Force - Displacement

LCD surface

Materials tensile test

Bending test at preset displacement

Bending test at preset displacement

LCD surface

Materials tensile test

Bending test at preset displacement

Bending test at preset displacement

Tester with thermostatic oven image*Jigs are not included. (Optional)

The MODEL-1840VCT is a large-size precision force tester designed for compression and tensile force tests up to 200 to 5000Kgf (2 to 50kN).
The test speed and control system conditions may be set on a touch panel.In combination with the MODEL-1016C amplifier, the basic
testparameters including the travel distance, repetition frequency and so forth may be set on the amplifier screen and measurement results are
displayed on a LCD.The stroke may be extended and the pitch of columns may be changed according to test sample sizes ofcustomers.Tensile/compressive tests in a specific environment may also be conducted.(Please contact us for a temperature range.)*Jigs are not included. (Optional)

The MODEL-1840VC is a large-size precision force tester designed for compression and tensile force tests up to 200 to 5000Kgf (2 to 50kN).
In combination with the MODEL-1016C amplifier, it displays measurement results on a LCD.It employs a ball screw feeding mechanism and a
servo motor for stable and wide-range test speed.The stroke may be extended and the pitch of columns may be changed according to test
sample sizes of customers.Tensile/compressive tests in a specific environment may also be conducted.(Please contact us for a temperature
range.)

Standard Specifications

Standard Specifications

1840VCT/200
1840VCT/500
1840VCT/1000
1840VCT/2000
1840VCT/5000
Model
2kN (200Kgf)
5kN (500Kgf)
10kN (1000Kgf)
20kN (2000Kgf)
50kN (5000Kgf)
Max. force
0.1 to 600mm/min.
0.1 to 300mm/min.
0.1 to 250mm/min.
Test speed
Free setting
Speed changing
700 mm1000 mm
950 mmStrokeProvided
Displacement display
Rotary encoder
Detector
0.01 mm
Display resolution
0.1 mm
Display accuracy±700.00 mm±1000.00 mm±950.00 mm
Max. display
Column interval
350 mm
400 mm
300 mm dia., center M20 x 1.5□290 mm M20 x 1.5
Table size
Servo motor
Drive motor
Driving mechanism
Ball screwLinear ball bush
Driving mechanism guideSize
W600xH1340xD580 mm
W600xH1714xD580 mm
W630xH1685xD610 mm
W720xH1714xD540 mm
WeightApprox.175kg
Approx.185kg
Approx.210kg
Approx.290kg
Approx.390kg
100VAC/200VAC, single-phase, 5A
100VAC/200VAC, single-phase, 10A
200VAC, single-phase, 10A 200VAC, three-phase, 10A
Power supply

1840VC/200
1840VC/500
1840VC/1000
1840VC/2000
1840VC/5000
Model
2kN (200Kgf)
5kN (500Kgf)
10kN (1000Kgf)
20kN (2000Kgf)
50kN (5000Kgf)
Max. force
1 to 600 mm/min.
1 to 300 mm/min.
Test speed
1 to 250 mm/min.
In five steps or continuous
Speed changing
700 mm1000 mm
950 mmStrokeProvided
Displacement display
Rotary encoder
Detector
Display resolution
0.01 mm
0.1 mm
Display accuracy±700.00 mm±1000.00 mm±950.00 mm
Max. display
Column interval
350 mm
400 mm
300 mm dia., center M20 x 1.5□290 mm M20 x 1.5
Table size
Servo motor
Drive motor
Driving mechanism
Ball screwLinear ball bush
Driving mechanism guide
Size
W600 x H1340 x D400 mm
W600 x H1714 x D400 mm W630 x H1655 x D410 mm W720 x H1714 x D540 mm
Approx.160 kg
Approx.170 kg
Approx.190 kg
Approx.270 kg
Approx.350 kg
Weight100VAC/200VAC, single-phase, 5A
100VAC/200VAC, single-phase, 10A
200VAC, single-phase, 10A 200VAC, three-phase, 10A
Power supply

Large Size Desk Top Type Force Testers

Large Size Force Testers
MODEL-1431VC/5000-20000

MODEL-1320VC/1321VC
Capacity: 10kN (1000 kgf)

MODEL-1322VC/1323VC
Capacity: 5kN (500 kgf)

Buckling test

Bending test

Materials tensile test

Capacity: 50 to 200 kN(5 to 20 tonf)
Force - Displacement

Bending test

Materials tensile test

MODEL-1324VC/1325VC
Capacity: 20kN (2000 kgf)

*Jigs are not included. (Optional)

These models are large size testers designed for compression and tensile force tests up to 500 to 2000Kgf (5 to 20kN).
They use the MODEL-1016C amplifier and display the measurement results on LCD panels.
They are available in force measurement only types and force-displacement measurement types, six types in all,whose
measurement force are 500Kgf, 1000Kgf and 2000Kgf.
With large test stands, 800mm long columns, and 360mm column interval, these are suitable to measurement of large parts.
The components and specifications may be changed according to test sample sizes of customers.

Standard Specifications

Standard Specifications
Model
Max.force
Test speed
Speed changing
StrokeDisplacement display
Detector
Display resolution
Display accuracy
Column length
Column interval
Table size
Drive motor
Driving mechanism
Driving mechanism guide
WeightSizePower supply

The MODEL-1431VC Series are multi-purpose large-size force testers designed for compression, tensile, bending andother tests of various large works at 1, 2, 5, 10 and 20 tonf. They employ precision ball screws and servo motors in the driving
mechanisms and rotary encoders for displacement detection.
They offer repetitive tests based on the max. value, breaking values, displacement and force, constant force tests,
step-force tests and so forth. They use the MODEL-1016C amplifiers.The test data and waveforms are displayed on LCD
panels.Various protection devices such as safety covers and area sensors are available according to customer's safety
requirements.

1320VC
1321VC
1322VC1323VC
10 KN (1000 kgf)
5 kN (500 kgf)
2 to 60 mm/min.
ln five steps or continuous
150mm
None
Provided
None
Provided
Rotary encoder
Rotary encoder

250 mm dia., center M20 x 1.5
DC brushless motor
Ball screw
Sliding contact
Approx. 120 kg
W510 x H1220 x D400 mm
100 to 240 VAC, single-phase, 5 A

1324VC
1325VC
20 KN (2000 kgf)
2 to 40 mm/min.

None

Provided
Rotary encoder

Model
Max.force
Test speed
Speed changing
StrokeDisplacement display
Detector
Display resolution
Display accuracy
Max. display
Column interval
Table size
Drive motor
Driving mechanism
Driving mechanism guide
WeightSizePower supply

1431VC/5000
5 tonf (50 kN)1000 mm

400 mm□ 290 mm, M20 x 1.5

Approx. 450 kg
W900 x H2345 x D600 mm
200 VAC, three-phase, 15 A

1431VC/20000
1431VC/10000
10 tonf (100 kN)
20 tonf (200 kN)
0.5 to 300 mm/minuteIn fifteen steps or continuous
800/1200 mm (not including grip)
Provided
Rotary encoder
0.01 mm
0.1 mm±1000.00 mm
520 mmServo motor
Ball screwLinear ball bush□ 400 mmApprox. 2000 kg
W1350 x H2535 x D1010 mm
200 VAC, three-phase, 20A*MODEL-1431VC/10000 and MODEL-1431VC/20000 are manufactured on a per-order basis to customer specifications.

650 mmHorizontal Force Tester

Switch Feeling Testers

MODEL-2152VCE
Capacity: 50N (5Kgf)
Force-Displacement

Capacity: 500N (50 kgf)
Force -Displacement
Feeing test of tact switches

Feeing test of key board

Feeing test of sheet switches

Max. force
Test speed
Speed changing
StrokeDisplacement display
Detector
Display resolution
Display accuracy
Z-axis stroke
Distance between test center & Z-axis plane
Table size
Drive motor
Driving mechanism
Driving mechanism guide
WeightSizePower supply

Z-axis (Test) Standard Specifications[Standard Configuration]●XYZ automatic stand●Measurement unit
●Load cell
●Measurement software

Max. force
Drive motor
Control motion
Drive range
Force measuring range
Force resolution
Force accuracy
Displacement measuring range
Displacement resolution
Displacement accuracy
Test speed range
Moving speed range
Analog output
Tester size
WeightPower supply

Standard Specifications

50N (5Kgf)
AC servo motor
UP/DOWN
0 to 200 mm
0 to 50N (0 to 5Kgf)
0.001N (1mN)
0.1N (100mN)
0 to 200 mm
0.001 mm±0.03 mm
0.1 to 50 mm/min.
1 to 15 mm/sec.
±10V (Force, displacement)
W750 x H900 x D560 mm

In five steps or continuous
150 mmProvided
Rotary encoder
0.01 mm
0.1 mm
40 mm
77.5 to 117.5 mm
W100 × D100 mm
DC brushless motor
Trapezoidal screwLinear ball bush
Approx.23kg
W518 × H314 × D345 mm
100 to 240 VAC,0.5A

Approx. 50Kg
100VAC※Personal computer is optional.

X-Y axes (Move) Standard Specifications

This tester is dedicated to feeling tests of force-displacement of various
switches, keyboard switches, dome switches, rubber switches, etc.
Three axes of X, Y and Z can be controlled automatically.
The tester is capable of testing 16 ON/OFF contacts maximum.
The tester is equipped with a jog dial to facilitate teaching.
Waveforms are displayed in real time on the amplifier and PC screen.

Drive motor
Control motion
Drive range
Displacement resolution
Displacement accuracy
Moving speed range
Specimen table size
Specimen carrying weight

Stepping motor
X: Right/left, Y: Front/back
X: 0 to 450 mm, Y: 0 to 200 mm
0.01 mm±0.1 mm
1 to 30 mm/sec.
W550xD280 mm

The MODEL-2152VCE is a horizontal force tester designed forcompression and tensile force tests up to 50 kgf (500 N) as well as
breaking tests and force -displacement correlation measurementof various electronic parts. It uses the Model-1016C amplifier anddisplays the measurement results on a LCD panel.

Standard Conﬁguration
Tester: MODEL-2152VCE
Measuring amplifier MODEL-1016C
Load cell:1pc

Small Size Electronic Parts Strength Evaluation Tester (IC Strength Tester)

Approx. 20Kg
Shearing jig

MODEL-1616W

Shearing direction

Capacity: 50N (5Kgf)
Force-Displacement

Epoxy resin

Die shearing test

Feeling test of tact switches

Standard Specifications
Max. force
Test speed
StrokeTest stroke

500N (50 kgf)
10 to 80 mm/min.

50N (5Kgf)
1 to 300 mm/min.
150 mm, ball screw
When using linear gauge : 0.000 to 30.000 mm

Standard Specifications
200N(20kgf)
Max. force
10 to 80 mm/min.
Test speed
In five steps or continuous
Speed changing
150 mmStrokeNone
Displacement display
40 mm
Z-axis stroke
Distance between test center & Z-axis plane 39 to 79 mm
W90 × D90 mm
Table size
DC brushless motor
Drive motor
Trapezoidal screw
Driving mechanism
Linearball bush
Driving mechanism guideApprox.23kg
WeightW518 × H228 × D345 mm
Size
100 to 240 VAC,single-phase,0.5A
Power supply

When using rotary encoder : 0.00 to 150.00 mm

This tester is dedicated to feeling tests of force-displacement of various
switches, dome switches, rubber switches, etc.
The Z axis is capable of measurement in the minimum unit of 1μ for
precision measurement.
The X axis can be moved to a desired position manually.
By using the CF card V-103A, feeling test points can be measured.

Displacement display
Detector
Right-left movement
Drive motor
Table size
WeightSizePower supply

30.000 mm, minimum 1μ possible（When using linear gauge）
Linear gauge (or Rotary encoder)
Manual: 200 mm
Servo motor
W480xD200 mm
Approx. 22Kg
W630xH910xD400 mm
100VACThe MODEL-2252RDH is designed to measure the adhesion
strength of electronic parts mounted on horizontal boards. The
sample fixing position may be adjusted in the Z direction. The
digital force gage RZ-20 is mounted on the force measuringside and the shearing jig is mounted at the top. It is also possible
to mount the hot plate (HTP-OP-1)on the sample fixing side.
The force measuring unit of this tester moves to the right and left.
It stores measured data,which may be printed or read in a PC.

Hot Plate for Thermel Load Tests (Common)(HTP-OP-1)
Specifications
Plate area:
60mm×60mm
Heating accuracy:
Normal temperature to 300℃℃
Temperature range:
Set temperature±10℃℃

Standard Conﬁguration
Tester: MODEL-2252RDH
RZ-20:1pc
Shearing jig:1pc(2mm,4mm or 8mm)

Options
Hot Plate
PrinterSmall Size Separation Evaluation Tester

Adhesion Tester

MODEL-1308UH

This is a low priced adhesion tester based on FTN1 - 13A
with only the mechanism changed. The maximum test speed
has been increased to 1000 mm/min to meet needs ofhigh-speed tests. In addition, a jig for 90-degree peeling thatmeets testing in accordance with JIS Z 0237 and software
FTN-2008P dedicated to adhesion test are also available.

FTN4-15A

Capacity: 200N (20 kgf)

Capacity: 50 N (5 kgf)
Force - Displacement
Resin/rubber
strength
Measuring
strengthof resin test
rubberTape weld
T-typepeeling
test
Measuring
separationof tape-fused
T brackets

The MODEL-1308UH employs a motor-driven gage mounting head, whichmoves up and down. The gage mounting head may be returned automaticallyby the action of a limit switch after completion of a test. The MODEL-1308UH
offers other various functions, including data output function. Refer to thecatalogs of the RZ series digital force gages. Use our various jigs andaccessories to extend the confines of tests. Refer to the pages of jigs and grips.

Standard Specifications
Max. force
Test speed
StrokeTable size
WeightSizePower supply

200 N (20 kgf)
10 to 300 mm/min.
400 mm
W250 x D140 mm
Approx. 20 kg
W255 x H810 x D300 mm
100 to 240 VAC,single-phase0.5A

90°separation test jig(Option installed)

Applicable Tests※Digital force gage is optional.

Separation tests at 90° and 180° (JIS-C 6481)
Various breakdown tests
Various non-destructive evaluation tests of parts
Penetration elasticity tests of fruits, etc.
Stress relief measurements of foods
Shearing tests
Bending tests

Embossed Carrier Tape Separation Tester
MODEL-2165P
Separation force: 5N (500 gf)

Standard Specifications
Peelstrength testof embossed
Embossedcarriertape tapeseparation strength test

The MODEL-2165P measures adhesion of various types oftapes. It conforms to the JIS and EIAJ Standards. The results
can be saved in a PC using the dedicated software. (Thissoftware is compatible with WindowsXP,7 or 8 & JapaneseOS.)Standard Specifications
Separation force
Resolution
Separation angle
Separation speed
Effective length
Applicable length
WeightSizePower supply

5 N (500 gf)
0.001 N (0.1 gf)
165 to 180 degrees(with five scales)
50, 100, 200, 300 & 400 mm/min.
400 mm88 mm
Approx. 16.5 kg
W630 x H300 x D260 mm
100 VAC / 220 VAC, 1A

Max. force
Test speed range
Moving speed range
Speed resolution
Resolution
Force
Displacement
Accuracy
Force

Optional jig
50 N (5 kgf)
5 to 1000 mm/min.
5 to 1000 mm/min. (inching & return)
0.1 mm/min.
0.01 N (50.00)
0.01 mm3000 Series: ±0.2% F.S.
UP Series: ±0.3% F.S.

Displacement

Effective stroke
Drive unit
Applicable load cell
Detector
Input/output
Digital input/output
Analog input/output
WeightSizePower supply±0.5 mm
100 mm (with 90-degree peeling jig installed)
Stepping motor
UP, M-3000 Series
Rotary encoder (2000 P/R)
USB (for connection to external PC)
Force (±10V)
Apporx. 35 kg
W300 x D400 x H820 mm
100 to 240 VAC, Single-phase, 3A

Applicable standard
Peeling method
Movable range (stroke)
Peeling point synchronizing method
Test plate
Specimen size
Safety measure

JIS Z 0237
90-degree peeling
100 mm (Standard: 85 mm)Moving amount synchronization with wire & pulley
50 x 125 x T2 or larger, SUS 304 (Surface roughness specified)
25 x 250
Safety cover on with & pulley and other areas

Software
Name
Function

FTN-2008P
Display of results, calculation (required numeric values, max.,
average, etc.), save, F-S graph display (real time, results, save data)

・Peak/bottom automatic acquisition: 50 maximum
・Peak/bottom display method: In the order of time or peak values
・Processing of multiple samples: (Simplified statistics)
・Set point measuring function: (JIS compatible)
・This software is compatible with Windows 7or 8 & Japanese or English OS.

Touch Panel Type Spring Testers

Touch Panel Type Spring Testers

MODEL-SHRⅢSeries

SHRⅢ-100

Capacity:1 kN（100 kgf）

SHRⅢ-1

Capacity:10 N（1 kgf）

SHRⅢ-200

Capacity:2 kN（200 kgf）

SHRⅢ-5

Capacity:50 N（5 kgf）

SHRⅢ-500

Capacity:5 kN（500 kgf）

SHRⅢ-10

Capacity:100 N（10 kgf）

SHRⅢ-1000

Capacity:10 kN（1 tonf）

SHRⅢ-50

Capacity:500 N（50 kgf）

SHRⅢ-2000

Capacity:20 kN（2 tonf）

Spring test

SHRⅢ-5000

Capacity:50 kN（5 tonf）

Outline of SHRⅢ Type Spring Tester
Test options
Screen display

Tensile test & compression test
Test data, totalization function, work No. registration(100 Nos.), & dispersion distribution bar graphKgf (N, Ib)
Unit options
Operation options Load measurement acc. to set length,measurement acc. to set load
Max. number of measurement stages: Ten
Max. stop time in each stage: 999.9 sec.

Input type for each
Teaching input, ten-key pad input, & step input
stage setting

Printing function Automatic, manual, Note input printing, & hard copy
Tolerance setting Two types selectable: ± and %5.7 TFT color LCD
Touch panel

The MODEL-SHR Ⅲ Series testers are the newest typehigh-performance spring testers. Their advanced
functions facilitate various measurement item setting andinput point setting, which have been very troublesome.
The user only has to follow instructions displayed on thescreen to input various items and may register amaximum of 100 work numbers. The screen provides
real-time display of measured data and dispersion
distribution bar graphs. It is possible to set a maximum often each measurement stages for both the normal spring
tests and length measurement according to preset loads.

Performances
Length resolution 0.01mm
Variable speed 1 to 600 mm/min.
Automatic distortion correction Corrects rigid distortion automatically
Automatic origin setting function Displays the data where the origin was set lastForce resolution 1/100000 max.
Extemal connection USB port
Protective functions Overload stop function, & emergency stop buttonStandard Specifications
Item
N force capability
N minimum reading
Unit of measurement
Test speed
Max. measurable length
Min. length reading
Compression plate diameter
Hook for tensile test
WeightSizePower supply

SHRⅢ-100

SHRⅢ-200

SHRⅢ-500

SHRⅢ-1000

SHRⅢ-2000

SHRⅢ-5000

1 kN（100kgf）

2 kN（200kgf）

5 kN（500kgf ）10 kN（1000kgf ）20 kN（2000kgf ）50 kN（5000kgf ）0.01 N（1gf）

1 N（100gf）

0.1 N（10gf）
kgf (N, Ib, inch for length measurement)
1 to 600 mm/min.
600 mm0.01 mm

200 mm dia

150 mm dia
1 stage

OptionApprox. 170kg

Approx. 190kg

Approx. 200kg

W500 x H1310 x D450 mm

W600 x H1372 x D540 mm

W600 x H1350 x D540 mm

Approx. 210kg

Approx. 290kg

Approx. 450kg
W900 x H2215 x D600 mm

100 VAC / 200 VAC

200 VAC

200 VAC, three-phase※The tensile spring jigs for the SHRⅢ-500 and larger force models are optional.

Software dedicated to SHRⅢ Series
Standard Specifications
Item
N force capability
N minimum reading
Unit of measurement
Test speed
Max. measurable length
Min. length reading
Compression plate diameter
Hook for tensile test
WeightSizePower supply

SHR Ⅲ-1

SHR Ⅲ-5

SHR Ⅲ-10

SHR Ⅲ-50

10 N（1kgf）

50 N（5kgf）

100 N（10kgf）

500 N（50kgf）

0.0001 N（0.01gf）

0.001 N（0.1gf）

0.001 N（0.1gf）

0.01 N（1gf）

kgf (N, Ib, inch for length measurement)
1 to 600 mm/min.
225 mm0.01 mm
60 mm dia.
1 stage

2 stages
Approx. 45kg
W450 x H565 x D345 mm
100 to 240 VAC

This is the software for dedicated SHRⅢ Series spring tester.
It acquires and sends real-time data to a PC and carries out
GO/NG judgment.
Measured data can be output in Excel format.

Specifications
Computer standard
Operatingsystem (OS)CPU
Main memory
Hard disk drive
Optical drive

DOS/V compliant IBM/PC compatible
Microsoft Windows 7 or 8; Japanese or English
Celeron processor 2.0 GHz min
1024 MB min. (2048 MB min. recommended)
20 to 200 MB according to operating environments
CD-ROM drive x1
When this software is installed.

DisplayHigh color (16 bits) min. (32 bits. min. recommended)
Desktop area 1024 x 768 min.

Torque Angle Testers

Torque Angle Testers

MODEL-5125VC/VCW (Wide range)

MODEL-5401VC-50/200

Capacity: 5N・m (0.5Kgf・m)
Torque-Angle
Twisting test of board with parts

Power window feeling measurement

Capacity: 5N・m/20N・m(0.5Kgf・m/2Kgf・m)
Torque-Angle

These are designed for complicated evaluation tests of various electronic,
mechanical and other parts, including the torque angle tests, sliding torque tests,
repetitive reducing torque measurement and so forth. The table in lower position
rotates. The span from the work is adjusted with the torque meter mount in upper
position, which is moved up and down manually. Select the proper torque meterand jig of the torque suitable to measurement. Test data and test waveform aredisplayed on the screen. In repetitive tests, waveforms are overwritten anddisplayed on the screen. Data are stored in the amplifier and printed.
5125VCStandard Specifications

5N・m (0.5Kgf・m)
0.1 to 20RPM

In five steps or continuous
0 to 340° or continuous
Provided
Rotary encoder
0.1 degree
1 degree

Capacity: 5N・m (0.5Kgf・cm)
Torque-Angle

Table size
Driving mechanism
Drive motor
WeightSizePower supply

110 mm

150 mm dia.
Table size
Ball speed reducer
Driving mechanism
Drive motor
DC brushlessServo motor
WeightApprox. 55Kg
Size
W350 x H767 x D475 mm
Power supply
100 to 240VAC, single-phase, 0.5A 100 to 240VAC, single-phase, 5Airnrvnmmr_

5401VC-50

5401VC-200

5N・m (0.5Kgf・m)

20N・m (2Kgf・m)
0.2 to 1RPM

0 to 340 degree
Provided
Rotary encoder
0.1 degree
1 degree
140 mmDistance between
rotational center & column

210 mmDistance between
rotational center & column

MODEL-5125VCT

ModelMax. output torque
Test speed
Angle setting
Displacement display
Detector
Display resolution
Display accuracy
Z-axis stroke

5125VCW (Wide range)

0.2 to 6RPM

Knob evaluation test

The MODEL-5401VC-50 and MODEL-5401VC-200 are capable of
torque angle measurement up to 340degrees. Samples are fixed in thelower positions, and the torque meter in the upper positions rotate. The
optional X-Y tables or jigs are installed on the sample measuring side.
Attaching the automatic centering table to the torque meter enables
smooth adjustment of the torque meter center to the work center. Thisfunction is convenient for samples that may not be rotated.

Standard Specifications
Model
Max. output torque
Test speed
Speed changing
Angle setting
Displacement display
Detector
Display resolution
Display accuracy
Z-axis stroke

Insertion & withdrawal test of key cylinder

120 mm
W350 x D320 mm
Pully transmission mechanism
DC brushless motor
Approx. 60Kg
W490 x H742 x D480 mm
100VAC, single-phase, 0.5A

100VAC, single-phase, 1A

Large Size Torque Testers

Aim-nnutlcmmnlriolnm_

ormwrrv

...M

-__

_

W1

clamwilll.=.~,...,.

Board evaluation test

Knob evaluation test

The MODEL-5125VCT designed for complicated evaluation tests of various
electronic, mechanical and other parts, including the torque angle tests,sliding
torque tests, repetitive reducing torque measurement and so forth. Various test
conditions may be set on a touch panel. The table in lower position rotates. The
span from the work is adjusted with the torque meter mount in upper position,
which is moved up and down manually. Select the proper torque meter and jig
of the torque suitable to measurement. Test data and test waveform aredisplayed on the screen. In repetitive tests, waveforms are overwritten anddisplayed on the screen. Data are stored in the amplifier and printed.

Standard Specifications
Max. output torque 5N・m (0.5Kgf・cm)
0.1 to 20RPM
Test speed
Free setting
Speed changing
0 to 340° or continuous
Angle setting
Displacement display Provided
Rotary encoder
Detector
0.1 degree
Display resolution
1 degree
Display accuracy

Z-axis stroke 210 mm
Distance between
rotational center & column

110 mm

150 mm dia.
Table size
Driving mechanism Ball speed reducer
Drive motor Servo motor
Approx. 55Kg
Weight
W350 x H767 x D475mm
Size
Power supply 100VAC, single-phase, 3A

MODEL-5127VC/500-5000

Capacity: 50N・m to 500N・m(500kgf・cm to 5000Kgf・cm)
Torque-Angle

The MODEL-5127VC Series are designed to measure correlation of the torque angles of large
automobile parts,electronic parts and so forth. Jigs or chucks are set on rotary tables in lower
positions. Spans from works are adjusted with the cross head in upper positions, which are
moved up and down by motors. Torque meters is fixed to the cross head. The MODEL-5127VC
Series employ the MODEL-1016C amplifier for versatile evaluation tests of parts, including
general torque breaking tests, repetitive durability evaluation tests and so forth.

Standard Specifications
Model
Max. output torque
Torque display
Load table rotation speed
Rotation accuracy
Rotational angle display & setting
Table size
Detector
Motor
WeightSizePower supply

5127VC/500

5127VC/2000

500Kgf・cm

2000Kgf・cm

5000Kgf・cm±500.0Kgf・cm±2000Kgf・cm±5000Kgf・cm

5127VC/5000

0.05 to 2RPM
±0.5%
±0 to 10000.0 degree
200 mm dia.

250 mm dia.

Rotary encoder
Servo motor
Approx.350Kg

Approx.400Kg
W900 x H2300 x D600 mm
200VAC, three-phase

Approx.700Kg

CUSTOM ORDER PRODUCTS

JIS-Rated Lead Free Solder Tester

Comprehensive Syringe Needle Force Measuring Tester

MODEL-1605 VC/NF

MODEL-1310VC/SL

Capacity: 50 N (5 kgf)
Force - Displacement

Capacity: 2kN (200Kgf)
Shearing test of solder joint of chip part

The MODEL-1310 is designed for comprehensive force tests ofvarious syringes shown below, for which various jigs are available. In
particular, a one-touch needle chuck is supplied for sharpness tests ofsyringe needles. Measured waveform and maximum value aredisplayed on the screen. Select the test items from those shown below.
Different jigs are used according to the test types. The MODEL-1310
has a vertically movable table.

45°pull test of QFP lead solder joint

The MODEL-1605 VC/NF is a precision tester designed for lead-free solder tests ofboards with parts in conformance to JIS Z 3198-6 and JIS Z 3198-7. Jigs are
replaced according to the test conditions. The pull point and shearing point are
adjusted with the X-Y table. Measured data are stored in the Model-1016C amplifier
and may be printed out or loaded into a PC after completion of measurement.

Standard Specifications

Standard Configuration

50 N (5 kgf)
Max. force
0.5 to 600 mm/min.
Test speed
250 mmStrokeDisplacement display Provided
Rotary encoder
Detector
Display resolution 0.01 mm
Display accuracy 0.1 mm■Tester: MODEL-1605VC■Amplifier: MODEL-1016C■Load cell: MODEL-3005 (5 kgf)
■Jig (Full set): M-500 FS

Displacement
reading

Capacity: 50 N (5 kgf)

Max. force
StrokeColumn length
Column interval
Test speed
Table size
WeightSizePower supply

Applicable Tests

10 μm (Standard)
1 μm (Option)

W475 x D150 mm, M1O x 1.5
Table size
Driving mechanism Ball screw
Servo motor
Drive motor
Approx. 50 kg
Weigh
W480 x H578 X D375 mm
Size
100 to 240 VAC, single-phase, 5A
Power supply

MODEL-1308U/NF

Standard Specifications

45° pull test of QFP leadsolder joint
JIS Z 3198-6
Shearing test of solder
joint of chip part
JIS Z 3198-7

The MODEL-1308U/NF is a low-price and compact
JIS-rated lead-free solder tester. It is designed for 45° pulltests of QFP lead solder and shearing tests of chip parts,
which are enabled by replacing the jigs and RZ series
digital force gages. Measured data are stored in the RZ
series digital force gage and may be printed out or loaded
into a PC after completion of measurement. The jigs arecommonly used for the MODEL-1605 VC/NF.

Applicable Tests
45° pull test of QFP lead solder joint: JIS Z 3198-6
Shearing test of solder joint of chip part: JIS Z 3198-7

Standard Specifications

Standard Configuration

Max. force 50 N (5 kgf)
Test speed 5 to 100 mm/min., variable
400 mmStrokeW250 x D140 mm
Table size
Approx. 20 kg
Weight
W255 x H810 X D300 mm
Size
Power supply 100 to 240 VAC, 0.5 A■Tester: MODEL-1308U■Digital force gage: select proper
RZ Series (up to 5 kgf)
■Jig (full set): M-500 FS

Torque feeling tester

2kN (200Kgf)
150mm
750mm
250mm
5 to 125mm/min.
200mm dia.
Approx. 41Kg
W356xH1102xD314mm
100 to 240VAC, single-phase 1A

Options
Select the following according to thetest requirements.

A. Sharpness measurement
B. Sharpness measurement in the condition where needle
bases are fixed to needles
C. Tensile strength measurement of needle-needle base adhesive
D. Slide strength measurement between barrels and pistons
E. Air tightness measurement between barrels and pistons
F.Dislocation (withdrawal strength) measurement ofneedles from barrels
G. Dislocation load measurement between needles and caps
H. Three-point bending measurement of needles only

Power Cable Bending Durability Tester (5CH)
MODEL-CBL/5S

This tester is designed to test a correlation between torque value and the rotation
angle. The acquired torque value and angle data are drawn on a torque-angle
graph. It is also possible to automatically acquire the click torque value and angle
during turning and show typical measurement values (click values).

This is designed for durability tests of cables in conformity to
the UL and JIS standards. It stops operation automaticallyat the set values (±90° & ±45°) and when the cable isbroken. (Interlock type and manual type are available.)

Switch & Silicone Rubber Durability Tester (3-CH)

Urethane Foam Load Measurings Instrument (Screen Display Type)MODEL-SR-3

The MODEL-SR-3 is designed for durability tests of various
electronic parts. It automatically stops a test at the preset cycle orwhen a test work is broken.

The MODEL-1900N allows all test patterns to be set on the touch
panel. It stores a maximum of ten JIS, JAS and any optional
patterns in the memory unit, which may be read out in tests. The
test data are displayed on the screen and may be printed out.

Grips & Jigs

Grips and Jigs
Standard Specifications
Capacity

Jaw Size(WxDmm)Jaw Max.
Opng (mm)

201

50N

20 x 15

8

※

203

2kN

25 x 2014

207-2K

2kN

25 x 2512※ M10 female

Approx.356

209

5kN

32 x 30

10

M20-P1.5 female

Approx. 1800

211

5kN

30 x 60

20

M20-P1.5 female

Approx. 2800

212

2kN

25 x 19

3.5

M10-P1.5 female

Approx. 545

220-50-45500N

50 x 45

0 - 45221-25

200N

W25

-

※ M6 female

Approx.115

221-50

2kN

W50

-

M10-P1.5

Approx. 376

MODEL

201
Small-size flat chuck

211

Cloth, metal & plastic

Large-size flat chuck

224-P-4／5
Pin vise

228G-10〜40／228H-10〜40
Film chuck
G: Rubber H: Sand paper

340-05／5(For torque)
Alignment table

CP-U-40／60／80
Round flat type compression table

203
Flat chuck212
Wedge-type chuck220-50-45
Small double side opening vise

225
Small loads of wires, etc.

230-45
Board mounting jig

340-L-5(For Tens. & Comp.)
Alignment table

HS-2〜8
Shear jig

MODEL-220

226-1／5／10／15
Pantograph chuck231
All-purpose type440
Z tableOJ-U-M6-M6 etc
Joint

Double Slide Vise

TR-1/4/5A/5
Chuck for round rod

207-2K
Flat chuck

Specifications

Max. open wicth 0 to 80 mm
Gripping force 1 kN
W90 x D30 mm
Blade size
Thread hole dia M6
2.1 kgWeight※Blades are optional
This vise has blades that open toward both sides from the center. Fix
it to the table of the instrument, and hold a test piece directly on thevise or fix the grip blades using the threaded holes in the upper pan.

X-Y Table
Model-330

209

Rubber, plastic, cloth & metal

Wedge-type chuck

221-25／50
Single roller chuck

Mountingscrew

Weight (g)M6 femaleApprox.125※ M10 female

Approx.442

Approx.958

Application

All-purpose typeType

Parallel ﬂat plate
individual screwing

Wedge type
Parallel ﬂat plate
individual screwing
Wedge type
Vice

Right-hand/left-hand thread

Film/cloth

Spring opening/closing

224-P-4/5

10N-

0- Φ 3.2※ M6 female

Approx.12

Round bar

For round bar/collet change type

225

0.5N

25 x 25

3

M6 femaleApprox.56

Gold wire

Clip type

226-1

5N

Tip jaw width 0.5-1

2

M6 female/M10 maleApprox.97

226-5

100N

Tip jaw width 52

M6 female/M10 maleApprox.100

226-10100N

Tip jaw width 10

2

M6 female/M10 maleApprox.100

226-15100N

Tip jaw width 15

2

M6 female/M10 maleApprox.107

227-20500N

20 x 104※

M6 femaleApprox.214

227-30500N

30 x 104※

M6 femaleApprox.233

228-10

20N

W10

1

M6 femaleApprox.37

228-20

20N

W20

1

M6 femaleApprox.49

228-30

20N

W30

1

M6 femaleApprox.64

228-40

20N

W40

1

M6 femaleApprox.77

230-45--

4-

Approx.850

231

200N

15 x 103

M6 femaleApprox.78

232

100N

Tip width 3xD8

1

M6 femaleApprox.30

340-05

0.5N・m---Approx. 315

340-5

20N・m

-

-

-

Approx. 1300

340-L-5

500N

-

-

-

Approx. 1500

Alignment table for Tens. & Comp.

440

200N

-

-

-

Approx. 3100

Z tableCP-U-40

100N

Φ 40

-

M6 femaleApprox.100

CP-U-60

100N

Φ 60

-

M6 femaleApprox.210

CP-U-80

100N

Φ 80

-

M6 femaleApprox.350

HS-2

30N

W2

-

M6 femaleApprox.44

HS-4

50N

W4

-

M6 femaleApprox.46

HS-6

50N

W6

-

M6 femaleApprox.48

Specifications

HS-8

50N

W8

-

M6 femaleApprox.49

1 mm/rotation
Travel±25 mm
X-axis travel±25 mm
Y-axis travel
2 kN
Max. force
Withstand tensile load 0.1 kN
55 x 60 mm
Table size
1 kg
WeightOJ-P-90---

M6 femaleApprox.748

TR-1

3kN

-

Φ 0.5-3

M6-P1 depth: 12

Approx. 70

TR-4

10kN

-

Φ 2-9

1/2-20UNF depth: 20

Approx. 500

TR-5A

50kN

-

Φ 3-13

TR-5

50kN

-

Φ 10-20227-20／30
All-purpose type232
All-purpose type

CB50-U3-D3
Bending jig

OJ-M10-F6,OJ-M10-M6 etc
Joint

Use this X-Y table together with AIKOH’s test stands for centeringor fine feeding of test pieces.

The screws with asterisks, ※ , need the OJ-U joints.
CP-U Series are not processed quenching※

1-12UNF depth40

Approx. 2500
Approx. 2500

Small parts
Pantograph typeAll-purpose typeThin plate

Screwing

Substrate holding
All-purpose typeOpening by springpressure, closing by screw

Alignment table for torque

Upper compression plate

Shear jig

Lower tension jig mounting plate

Round bar

Scroll type

Indicators
MODEL-0218B

MODEL-0215T

Load cells
This is a low priced simple digital display meter.
This compact and light-weight equipment has the following features:

Features

Standard specifications

1. Data hold function (Sample hold)
2. Peak hold function
3. External reset function
4. One-touch auto zero function
5. High and low values settable as desired
6. Analog output

Display range
Response
Sampling rate
Decimal point
Temperature drift
Applied voltage
Power supply

99999-19999
10, 100, 1kHz
15times/sec.
Free setting
0.02% F.S.%/℃
5VDC
100V, 110V, 200V or 220V

This is a digital display meter compliant with the TEDS Standard (IEEE1451.4Class 2 mixed mode interface).
When combined with various TEDS-compatible strain gauge type sensors,sensitivity can be calibrated easily and accurately.
Zero calibration, span calibration, upper/lower limit comparison, digital/analog
filtering, motion detection and zero tracking can be set manually on the display.

Features●●

What is "TEDS"?

●"TEDS" stands for "Transducer Electronic Data
Sheet".When a memory containing TEDS dataof a sensor is installed, the information of thesensor can easily be transferred to a display
meter having a function of reading thememory.
Usable load cells (made by AIKOH)
MODEL-CM, UP, QF, DCD, CH, US, DUD, CB Series

●
●
●
●
●

The use of the TEDS function ensures easy and accurate sensitivity calibration.
The display meter needs not be used with a sensor as a pair,but their combination may be changed as desired.
A sensor in which no TEDS data has been written can also be used.
Static strain can be measured.
The conversion cable for connection of MODEL-0215T is a standard accesorry.
CE mark-compliant.
The power supply is 100 to 240VAC.
Optionally, BCD parallel data output, RS232C interface and D/A converter(Voltage/current output) are available.

MODEL-QF
MODEL-CK
MODEL-VCV
MODEL-US
Small size load cell for compression test High performance load cell for compression test Load cell for tension/compression Non-Rotary Type Torque Meter
Capacity: 100N to 20kN
Capacity: 500N to 100kN
Capacity: 10kN to 500kN Capacity: 0.2 N・m to 1kN・m●Standard Specifications
Rated output: 1mV/V (10 & 20kN: 1.5mV/V)
Non-linearity: 1%R.O. (10 & 20kN: 2%R.O.)
Hysteresis: 1%R.O. (10 & 20kN: 2%R.O.)
Repeatability: 1%R.O.
Recommend applied voltage: 5V
Allowable applied voltage: 7V
Temp. Compensation range:-5 to +65 ℃(2 to 20kN: -5 to +50 ℃)
Allowable compensation range:-10 to +70 ℃ (2 to 20kN: -10 to +60 ℃)
Allowable overload: 120%R.C.

●Standard Specifications
Rated output: 2mV/V±0.25%(50 & 100kN: 2mV/V±0.5%)
Non-linearity: 0.15%R.O.
(50 & 100kN: 0.1%R.O.)
Hysteresis: 1%R.O.
Repeatability: 0.05%R.O.
Recommend applied voltage: 8V
Allowable applied voltage: 12V
Temp. Compensation range: -10 to +70 ℃
Allowable compensation range: -15 to +80 ℃(50 to 100kN: -15 to +75 ℃)
Allowable overload: 150%R.C.

●Standard Specifications
Rated output: 1mV/V
Non-linearity: 1%R.O.
Hysteresis: 1%R.O.
Repeatability: 0.3%R.O.
Recommend applied voltage: 6V
Allowable applied voltage: 10V
Temp. Compensation range: -10 to +70 ℃
Allowable compensation range: -20 to +80 ℃
Allowable overload: 150%R.C.

●Standard Specifications
Rated output: 2mV/V±0.5%
Non-linearity: 0.15%R.O.
Hysteresis: 0.1%R.O.
Repeatability: 0.1%R.O.
Recommend applied voltage: 10V
Allowable applied voltage: 15V
Temp. Compensation range: -10 to +70 ℃
Allowable compensation range: -20 to +80 ℃
Allowable overload: 150%R.C.

MODEL-3000 Series

Equivalent input/TEDS

10VDC, 2.5V±10%, current 120mA
0 to ±3.0mV/V

Calibrating range
Calibrating accuracy

Zero adjustment range
Minimum input sensitivity
Non-linearity
A/D conversion speed
Analog filter
Analog output
TEDS function
Response speed
Peak-hold function(analog and digital hold system) Accuracy
When reset
Display
Display range
Power supply
Operating temperature range
Operating humidity range
External dimensions
WeightAccessory
Option0.3 to 3.0mV/V
0 to ±2.0mV/V
Within 0.02% F.S. (When input is 3mV/V or over.)100times/sec.
4, 10, 100Hz (Default), 3kHz (Set on the panel)
Voltage: 2V±5% max. per 1mV/V, force registance: 2kΩ or more
Response frequency: Approx. 5kHz/-3dB (Not passing through analog filter)
IEEE1451.4 Class2, mixed mode interface
Approx. 1kHz (Waveform width 2ms: 3mV/V input, analog filter 3kHz)
0.2% F.S. or better
50μs max.
±19999100 to 240VAC, Approx. 7W-10 to +40℃ (Storage temperature range: -40 to +80℃)
85% RH max. (No condensation)
Approx. W96 x H96 x D146mm (Projections are not included)
Approx. 1kg
AC power cable, 1piece, TEDS conversion cable 1piece, instruction manual (CD-ROM) 1copy
BCD parallel data output, RS232C interface
D/A converter (Voltage/current output)●Standard Specifications
Rated output: 1mV/V±1%
Non-linearity: 0.5%R.O.
Hysteresis: 0.5%R.O.
Repeatability: 0.1%R.O.
Recommend applied voltage: 10V
Allowable applied voltage: 15V
Temp. Compensation range: -10 to +70 ℃
Allowable compensation range: -20 to +80 ℃
Allowable overload: 150%R.C.

●Standard Specifications
Rated output: 2mV/V±1%
Non-linearity: 0.15%R.O.
Hysteresis: 0.1%R.O.
Repeatability: 0.1%R.O.
Recommend applied voltage: 10V
Allowable applied voltage: 15V
Temp. Compensation range: -10 to +70 ℃
Allowable compensation range: -20 to +80 ℃
Allowable overload: 150%R.C.

MODEL-CB
Load beam
Capacity: 50N to 10kN

●Standard Specifications
Rated output: 1.5mV/V±1%(10 to 50N・m: 1mV/V)
Non-linearity: 0.3%R.O.
Hysteresis: 0.2%R.O.
Repeatability: 0.3%R.O.
Recommend applied voltage: 10V
Allowable applied voltage: 15V
Temp. Compensation range: -10 to +70 ℃
Allowable compensation range: -20 to +75 ℃
Allowable overload: 150%R.C.

●Standard Specifications
Rated output: 2mV/V±1%
Non-linearity: 0.05%R.O.
Hysteresis: 0.05%R.O.
Repeatability: 0.05%R.O.
Recommend applied voltage: 10V
Allowable applied voltage: 15V
Temp. Compensation range: -10 to +70 ℃
Allowable compensation range: -20 to +80 ℃
Allowable overload: 150%R.C.

Tensile & Compression Type Load Cells
While these load cells with female threads on both sides are designed for both tensile andcompression tests, they feature high precision, high output, and less output errors inpushing and pulling, resulting in high reliability. They are widely used for performance tests
and industrial measurement of materials and car parts in as wide ranges as 50 N to 20 kN.

Dimensions

0.2% F.S. or better, provided that the sensor sensitivity it set to 0.5mV/V min.
1μV/count (1/10000 guaranteed at input above 1mV/V)●Standard Specifications
Rated output: 1mV/V±1%
Non-linearity: 0.3%R.O.
Hysteresis: 0.3%R.O.
Repeatability: 0.2%R.O.
Recommend applied voltage:
10V (6V at 1N・m or less)
Allowable applied voltage:
15V (10V at 1N・m or less)
Temp. Compensation range: -10 to +70 ℃
Allowable compensation range: -20 to +80 ℃
Allowable overload: 150%R.C.

MODEL-CM
MODEL-DCD
MODEL-DUD
MODEL-QR
Very small size load cell for compression test High performance load cell for compression test Small high performance load cell for tension/compression test Rotary type torque meter
Capacity: 10N to 2kN
Capacity: 500N to 200kN
Capacity: 500N to 20kN
Capacity: 10N・m to 2kN・m

Specifications
Applied voltage
Signal input range●Standard Specifications
Rated output: 2mV/V±0.25%(10kN: 1.5mV/V)
Non-linearity: 0.1%R.O.
Hysteresis: 0.1%R.O.
Repeatability: 0.05%R.O.
Recommend applied voltage: 10V
Allowable applied voltage: 15V
Temp. Compensation range: -10 to +70 ℃
Allowable compensation range: -20 to +80 ℃
Allowable overload: 150%R.C.

MODEL-CH
Load cell for compression test
Capacity: 5kN to 500kN

Standard Specifications
Rated capacity 20 N to 20 kN
Rated output 2mV/V=1%(20N:1mV/V,10kN & 20kN:1.8mV/V)
Non-linearity 0.1% R.O.
0.1% R.O.
Hysteresis
Repeatability 0.1% R.O.
Recommended applied voltage 10 V
Allowable applied voltage 15 V
I/O resistance 350 Ω ±2%
Temp.compensation range -10 to +70℃
Allowable compensation range -20 to +80℃
Temp.influence upon zero point ±0.005% R.O. / ℃
Temp.influence upon output ±0.005% / ℃
Allowable overload. 150% R.C.

Model
3005
3020
3050
3200
3500
3800
3900

Rated capacity ФA ФB
50 N
60 50
200 N
60 50
500 N
60 50
2KN
60 50
5KN
68 58
10KN
60 50
20KN
60 50

G
C D F
2 5 80 M10 P1.5
2 5 80 M10 P1.5
2 5 80 M10 P1.5
2 5 80 M10 P1.5
2 10 90 M20 P1.5
3 15 120 M20 P1.5
3 15 120 M20 P1.5

H
20
20
20
20
36
36
36

3000 Series

I
17
17
17
17
30
30
30

※3002 (20 N) type is also available. The rated output is 1 mV/V.

UP Series
3.4 coreshielded cable

4-3.5 hole

5N, 20N
Rated capacity
Allowable overload 150%R.C.
Hysteresis 0.2%R.O.
Temp.compensation range -10~＋70 ℃

Rated output
Non-linearity
Repetition Performance
Allowable compensation range

1 mV/V
0.2%R.O.
0.1 %R.O.
-20~＋80 ℃

Standard Softwares

Standard Softwares

FS-700 (MODEL-1016 Series Basic Dedicated Software)

RZ-3000(Software dedicated to RZ Series)

This is data processing application software dedicated to
MODEL-1016 series amplifiers. Real-time data output andwaveform drawing at high-speed sampling is possible. For tests,
almost all conditions of MODEL-1016 series can be set on a PC.
Not only continuous testing, but the X.LS format and CSV format
to directly save data to Excel are possible. The available printstyles include overlapped, shifted and split print. (The PC and
Excel are to be provided by the user.)

Recommended Environment
Japanese Windows7or8
Memory: 1G bytes or more
38400bps serial port: 1
CD-ROM drive for program installation: 1
Internet Explorer 5.0 or later
English OS accepted
MODEL-1016 Series Basic Version onlyThis is dedicated software designed to store data acquired by RZ
Series in a PC.
It acquires and sends real-time data to a PC to show waveformgraphs and maximum values.
Measurement data can be output in CSV files and output CSV
files can be opened using Excel.
The standard interface is the USB interface. (The USB cable isincluded with the RZ force gage as a standard accessory.)

Recommended Environment
Windows7or8, Japanese or English
Memory: 2G bytes min.
CD-ROM drive for program installation x1

FL-2005 (Software exclusive for the RX-FL Series)

FSN-500 (V-103/V-103A Switch Feeling Dedicated Software)This is application software dedicated to V-103 and V-103A
switch feeling. When the optional ROM card V-103 or CF card
V-103A is used, measured data/graph data can be loaded to a
PC and processed, saved and printed. (The PC and Excel are to
be provided by the user.)

Recommended Environment
Japanese Windows7or8
Memory: 1G bytes or more
38400bps serial port: 1
CD-ROM drive for program installation: 1
Internet Explorer 5.0 or later
English OS accepted

The FL-2005 is data loading application software designed
exclusively for the RX-FL Series. It runs on Microsoft Windows
operating systems. It receives feeling measurement data output
by the RX-Fl Series, which is connected to the RS232C port of a
PC with RX-OP-2, carries out GO/NG judgment, data recording
and other processing. It has various functions: Judgment,
Various types of totalization processing, Printing, Simple
histogram display, Data storage in CSV files, etc.

Recommended Environment
Japanese WindowsXP only
Memory: 1G bytes or more
38400bps serial port: 1
CD-ROM drive for program installation: 1
Internet Explorer 5.0 or later