R

User Guide
Elcometer 311
Automotive Refinishing Gauge

www.elcometer.com

R

CONTENTS
SectionPage

1

Gauge Overview

en-2

2

Box Contents

en-2

3

Using the Gauge

en-3

4

Getting Started

en-3

5

Taking a Reading

en-5

6

Statistics - Model T

en-6

7

Calibrating the Gauge

en-6

8

Downloading Data - Model T

en-7

9

Upgrading your Gauge - Model T

en-8

10

Spares & Accessories

en-8

11

Warranty Statement

en-8

12

Technical Specification

en-9

13

Legal Notices & Regulatory Information

en-10

14

Appendix 1: Summary of Gauge Icons

en-11

15

Appendix 2: ElcoMaster® Collect Design

en-12

Pb

Hg

For the avoidance of doubt, please refer to the original English language version.
Gauge Dimensions: 141 x 73 x 37mm (5.55 x 2.87 x 1.46")
Gauge Weight:
156g (5.5oz) including batteries
Applicable Patents:
FNF Gauges: US5886522; Model T Gauges: US7606671© Elcometer Limited 2015 - 2024. All rights reserved. No part of this document may be reproduced, transmitted,
transcribed, stored (in a retrieval system or otherwise) or translated into any language, in any form or by anymeans (electronic, mechanical, magnetic, optical, manual or otherwise) without the prior written permission of
Elcometer Limited.

en-1

www.elcometer.com

R

1 GAUGE OVERVIEW
6

1
2
7
3
8

4

5

1
2
3
4
5
6
7
8

LED Indicators - Red (left), Green (right)
Colour Screen
Multifunction Softkeys
On/Off Key
Integral Probe
USB Data Output Socket (below cover)
Battery Compartment (¼ turn open/close)
Wrist Strap Connection

2 BOX CONTENTS
<
<
<
<
<
<
<
<

a

Elcometer 311 Automotive Refinishing Gauge
Calibration Check Piece; Steel & Aluminium (FNF gauges only)
Calibration Check Foil; 125µm (5mils)a
Wrist Strap
Protective Case
2 x AA Batteries
Test Certificate
User Guide

Nominal Value

www.elcometer.com

en-2

R

3 USING THE GAUGEaPower: Batteries or USB -ahincluding battery life indicatorbbBluetooth On (Model T) -c

Grey: not paired; Orange: pairedc

Substrate Type - F: Ferrous; FNF: Ferrous;
F N F: Non-FerrousdUnits of Measurement - µm, milse

Menu SoftkeyfStatistics Softkey (Model T) -

d

e
g

f

see page en-6g

Calibration SoftkeyhStatistics (Model T)

4 GETTING STARTED
4.1 ENSURING YOUR GAUGE HAS THE LATEST FIRMWARE& UPGRADING YOUR GAUGE - MODEL T
To ensure that your gauge has the most up-to-date gauge firmware,
allowing you to benefit from the latest features and functionality, we
recommend that the gauge is connected to ElcoMaster® on a regularbasis and before first use.
Simply connect the gauge via USB to an internet connected computerrunning ElcoMaster® using the ‘Connect Gauge’ feature. If a later
version of the gauge firmware is available, ‘Update Gauge’ will be
displayed to the right of the gauge details. Click ‘Update Gauge’ toinstall the latest firmware.
4.2 FITTING THE BATTERIES
Each gauge is supplied with 2 x AA alkaline
batteries.

+

To insert or replace the batteries:
1
Lift the latch on the battery compartment
cover and rotate anti-clockwise to remove
the cover.
+2
Insert 2 batteries taking care to ensure
correct polarity.
3
Refit the cover and rotate the latch clockwise to close.
en-3

www.elcometer.com

R

4 GETTING STARTED (continued)
The battery condition is indicated by a symbol in the top right of thedisplay ():
4 Full symbol (orange) = batteries at full capacity
4 Empty symbol (red, flashing) = batteries at lowest

sustainable level
4.3 SWITCHING ON / OFF
To switch on: Press and hold the ON/OFF button until the Elcometerlogo is displayed.
To switch off: Press and hold the ON/OFF button until the screen
goes blank.
Note: The gauge will switch off automatically after 1 minute of inactivity.

4.4 SETTING THE BEEP VOLUME
The gauge beeps each time a measurement is taken. To set the beep
volume:1
2
3

Press the ‘ ’ softkey.
Use the éê softkeys to highlight ‘ ’ and press ‘ ’ to select.
Use the éê softkeys to adjust the volume and press ‘ ’ to set or‘ ’ to disregard any changes made and return to the previous
screen.

4.5 SETTING THE SCREEN BRIGHTNESSThe screen brightness can be set to ‘ ’ (Auto) - the brightness isadjusted automatically using the gauge’s ambient light sensor - or canbe set by the user to suit the measurement conditions. To set thescreen brightness:
1
2
3

Press the ‘ ’ softkey.
Use the éê softkeys to highlight ‘ ’ and press ‘ ’ to select.
Use the éê softkeys to adjust the screen brightness and press‘ ’ to set or ‘ ’ to disregard any changes made and return to theprevious screen.
4 The default setting is ‘ ’ (Auto).

www.elcometer.com

en-4

R

4 GETTING STARTED (continued)
4.6 SELECTING THE MEASUREMENT UNITSThe Elcometer 311 can take measurements in µm and mils. To set themeasurement units:
1
2

Press the ‘ ’ softkey.
Use the éê softkeys to highlight ‘µm/mil’ and press ‘ ’ toalternate between ‘µm’ and ‘mil’, followed by the ç softkey toreturn to the main reading screen.

5 TAKING A READING
Before taking a reading, it is good practice to check the performanceof the gauge using the calibration check procedure, see Section 7.2‘Checking the Gauge Calibration’ on page en-6.
To take a reading:1
Whilst holding the gauge perpendicular, gently place the probe
onto the surface.
4 The reading may be inaccurate if the probe is not held flat against thesurface.

2

The coating thickness is shown on the display.
4 ‘>500µm’ (‘>20mils’) indicates a reading outside the gauge range.

3

For subsequent readings, lift the probe off and then replace it on
to the coated surface.

Using ElcoMaster® the Elcometer 311 Model T can simultaneouslytransmit the live reading to a PC or mobile device for report generation,see Section 8 ‘Downloading Data’ on page en-7 for more information.
The Elcometer 311 Model T calculates and displays statistical values
as each reading is taken, see Section 6 ‘Statistics’ on page en-6 fordetails.

en-5

www.elcometer.com

R

6 STATISTICS - MODEL T
The Elcometer 311 Model T calculates and displays the following
statistical values as each measurement is taken:
ŸŸŸŸ

Number of readings (n:)
Average reading (X:)
Lowest reading (Lo:)
Elcometer Index Value (EIV:)
Used to assess a coating’s overall quality. Ideal for use in the automotive refinishingindustry. Patent number US7606671.

To display or hide the statistics from the screen, press the ‘softkey.

’

To clear the current statistical values, press and hold the ‘’ softkey
for approximately 3 seconds (when statistics are displayed). The
gauge will beep and the statistics will reset to zero.
7 CALIBRATING THE GAUGE
7.1 USING ZERO CALIBRATIONThe user can calibrate the gauge using zero calibration, a simple
method for calibrating on smooth substrates.
1
2

Press the ‘ ’ softkey.
When prompted, whilst holding the gauge perpendicular, gently
place the probe onto the uncoated metal substrate. ‘Zero’ will be
displayed and the gauge will return to the main reading screen.

Note: FNF gauges must be calibrated in both the ferrous and non-ferrous mode
to ensure accuracy of readings.

7.2 CHECKING THE GAUGE CALIBRATIONThe user can check the calibration and performance of the gauge
using the calibration check foil and piece(s) supplied.
1
2
3

Place the calibration check foil on to the check piece.
Whilst holding the gauge perpendicular, gently place the probe
onto the foil.
The gauge should read the value printed on the foil label ±25µm(1mil).

If the gauge does not display a valid reading, it may requirere-calibration - see Section 7.1.
www.elcometer.com

en-6

R

8 DOWNLOADING DATA - MODEL T
8.1 USING ELCOMASTER® ON A PC
Using ElcoMaster® - available as a free download at elcometer.com the Elcometer 311 Model T can simultaneously transmit the live
reading via USB or Bluetooth®, to a PC for report generation. For
more information on ElcoMaster® visit www.elcometer.com.
8.2 USING ELCOMASTER® MOBILE APPS
Ideal when out in the field or on-site, using the ElcoMaster® Mobile App
users can:ŸŸŸStore live readings directly on to a mobile device and save theminto batches together with GPS coordinates.
Add photographs of the test surface (via the mobile device’s camera).
Map readings on to a photograph or diagram (when using the‘Collect Design’ function in ElcoMaster® - see Appendix 2 ‘ElcoMaster®
Collect Design’ on page en-12)ŸInspection data can be transferred from mobile to PC for further
analysis and reporting.

For more information on ElcoMaster® Mobile Apps visit
www.elcometer.com.

en-7

www.elcometer.com

R

9 UPGRADING YOUR GAUGE - MODEL T
Gauge firmware can be upgraded to the latest version by the user via
ElcoMaster®, as it becomes available. ElcoMaster® will inform the userof any updates when the gauge is connected to the PC with an internet
connection.
To upgrade the Elcometer 311 Model B firmware, please contact yournearest Elcometer Distributor or Service Centre.
10 SPARES & ACCESSORIESThe following spares and accessories are available from your local
Elcometer supplier or direct from Elcometer.
Description
Part Number
Calibration Check Piece; Steel (Ferrous)
T99916925
Calibration Check Piece; Aluminium (Non-Ferrous) T99916901
Calibration Check Foil; 125µm (5mils)a
T99022570-7A
USB Cable
T99921325
Self Adhesive Screen Protectors, Pack of 10
T99922341
11 WARRANTY STATEMENT
Elcometer 311 gauges are supplied with a one year warranty against
manufacturing defects, excluding contamination and wear.
The warranty can be extended to two years within 60 days of purchase
via www.elcometer.com.

aNominal Value

www.elcometer.com

en-8

R

12 TECHNICAL SPECIFICATION
Range

0 - 500µm (0 - 20mils)

Accuracyb±5% or ±20µm (1.0mil)

Resolution

10µm (0.5mil)

Minimum Substrate
Thicknessc
Measurement Speed
OperatingTemperature
Relative Humidity

Steel:
Aluminium:

800µm (30mils)
300µm (12mils) - FNF gauges only

Up to 60 readings per minute-10 to 50°C (14 to 122°F)
0 to 95%

Power Supply

2 x AA Batteries or via USB Cable

Battery Lifed

Alkaline: Approximately 16 hours
Lithium: Approximately 24 hours

Gauge Dimensions
Gauge Weight(including batteries)(rechargeable batteries can also be used)

14.1 x 7.30 x 3.70cm(5.55 x 2.87 x 1.46")
156g (5.5oz)

Can be used in accordance with:ISO 2178, ISO 2808, ISO 2808-7C, ISO 2808-7D, ISO 2808-12A,
ISO 2808-11B, ASTM E376, JIS K 5600-1-7, AS/NZS 1580.108.1

bWhichever is the greater.
For specified accuracy.
dRechargeable batteries may differ.
c

en-9

www.elcometer.com

R

13 LEGAL NOTICES & REGULATORY INFORMATION
Declaration of Conformity CE:
Elcometer 311 Model B complies with the requirements of the following EU Directives:
2014/30/EU
EMC Directive
2011/65/EU
Restriction of the Use of certain Hazardous Substances in Electrical andElectronic Equipment (RoHS) Directive
https://downloads.elcometer.com/Declaration_of_Conformity/English/DoC_311C_B.pdf
Declaration of Conformity UKCA
Elcometer 311 Model B complies with the requirements of the following UK Standards:
S.I. 2016 No. 1091
Electromagnetic Compatibility Regulations 2016
S.I. 2012 No. 3032
Restriction of the Use of Certain Hazardous Substances in Electrical andElectronic Equipment (RoHS) Regulations 2012https://downloads.elcometer.com/Declaration_of_Conformity/English/DoC_311C_B_UKCA.pdf
Declaration of Conformity CE:
Elcometer 311 Model T complies with the requirements of the following EU Directives:
2014/53/EU
Radio Equipment Directive
2014/30/EU
EMC Directive
2011/65/EU
Restriction of the Use of Certain Hazardous Substances in Electrical andElectronic Equipment (RoHS) Directive
https://downloads.elcometer.com/Declaration_of_Conformity/English/DoC_311C_T.pdf
Declaration of Conformity UKCA:
Elcometer 311 Model T complies with the requirements of the following UK Standards:
S.I. 2017 No. 1206
Radio Equipment Regulations 2017
S.I. 2016 No. 1091
Electromagnetic Compatibility Regulations 2016
S.I. 2012 No. 3032
Restriction of the Use of Certain Hazardous Substances in Electrical andElectronic Equipment (RoHS) Regulations 2012https://downloads.elcometer.com/Declaration_of_Conformity/English/DoC_311C_T_UKCA.pdf
Operational Frequency Band: 2,402 - 2,480 MHz
Maximum Transmitted Power: <4 dBm
This product is Class B, Group 1 ISM equipment according to CISPR 11.
Class B product: Suitable for use in domestic establishments and in establishments directly connected to a low voltage power supply
network which supplies buildings used for domestic purposes.
Group 1 ISM product: A product in which there is intentionally generated and/or used conductively coupled radio frequency energy
which is necessary for the internal functioning of the equipment itself.
The USB is for data transfer only and is not to be connected to the mains via a USB mains adapter.
This device complies with Part 15 of the FCC Rules. Operation is subject to the following two conditions: (1) this device may notcause harmful interference, and (2) this device must accept any interference received, including interference that may cause
undesired operation.
Elcometer 311 Model T: The FCC ID and Bluetooth SIG QDID can be accessed by pressing ‘ ’ followed by ‘ ’.
Elcometer 311 Model B: The regulatory information can be accessed by pressing and holding the ON/OFF button whilst the gauge isswitched off until the menu is displayed, followed by ‘ ’.
NOTE: This equipment has been tested and found to comply with the limits for a Class B digital device, pursuant to Part 15 of the
FCC Rules. These limits are designed to provide reasonable protection against harmful interference in a residential installation. Thisequipment generates, uses and can radiate radio frequency energy and, if not installed and used in accordance with the instructions,
may cause harmful interference to radio communications. However, there is no guarantee that interference will not occur in aparticular installation. If this equipment does cause harmful interference to radio or television reception, which can be determined byturning the equipment off and on, the user is encouraged to try to correct the interference by one or more of the following measures:-- Reorient or relocate the receiving antenna.
-- Increase the separation between the equipment and receiver.
-- Connect the equipment into an outlet on a circuit different from that to which the receiver is connected-- Consult the dealer or an experienced radio/TV technician for help.
To satisfy FCC RF Exposure requirements for mobile and base station transmission devices, a separation distance of 20 cm or moreshould be maintained between the antenna of this device and persons during operation. To ensure compliance, operation at closer
than this distance is not recommended. The antenna(s) used for this transmitter must not be co-located or operating in conjunction
with any other antenna or transmitter.
Modifications not expressly approved by Elcometer Limited could void the user’s authority to operate the equipment under FCC rules.
This device complies with Industry Canada license exempt RSS standard(s). Operation is subject to the following two conditions: (1)this device may not cause interference, and (2) this device must accept any interference, including interference that may cause
undesired operation of the device.
This Class B digital apparatus complies with CAN ICES-3 (B)/NMB-3(B).
and ElcoMaster® are registered trademarks of Elcometer Limited, Edge Lane, Manchester,M43 6BU. United Kingdom
are trademarks owned by Bluetooth SIG Inc and licensed to Elcometer Limited.
.All other trademarks acknowledged.
Head-Office: Elcometer Limited, Edge Lane, Manchester, M43 6BU, United Kingdom.
Rwww.elcometer.com

en-10

R

14 APPENDIX 1: SUMMARY OF GAUGE ICONS
Icon

Description

ModelPower: Batteries - Including Battery Life Indicator

B, T

Power: USB

B, T

Bluetooth On - Grey: not paired; Orange: paired

T

F,
Substrate Type - F; Ferrous (Model B)
F NF, F N F FNF Gauges (Auto Detect) - FNF: Ferrous; FNF: Non-Ferrous

B, T

µm

Units of Measurement - µm

B, T

mils

Units of Measurement - mils

B, T

Menu Softkey

B, T

Statistics Softkey

T

Calibration Softkey

B, T

Units of Measurement Menu Option

B, T

Volume Setting Menu Option

B, T

Screen Brightness Menu Option

B, T

Legal Information Menu Option

T

Select / Set Softkey

B, T

Escape Softkey

B, T

Elcometer Contact Details Menu Option

T

en-11

www.elcometer.com

R

15 APPENDIX 2: ELCOMASTER® COLLECT DESIGNThe ‘Collect Design’ function in
ElcoMaster® allows users to set up acollection template to guide Inspectors
where, and in which order, readings
should be taken.
This image can be used as a collection
guide in ElcoMaster® on a PC or can betransferred to a mobile device using the
ElcoMaster® Mobile App.
When creating a new ‘Collect Design’ the user:
Ÿ
Ÿ
Ÿ
Ÿ
Ÿ

Adds the required image to collect readings against;
Specifies the number of readings to be taken;
Enters any relevant limits;
Can request a prompt to begin the next set of readings or begin
automatically;
Choose to create a new batch for each set of readings or create asingle batch with multiple readings.

To set up a new ‘Collect Design’:1
Switch ElcoMaster® to ‘Advanced
Mode’ by clicking the expand arrowand selecting ‘Advanced Mode’.
2
Click on ‘Library’à’New’à’Collect
Design’.
3
Follow the on-screen prompts andclick ‘ Finish’ to begin the design.
4
To indicate on the image where
each reading is to be taken and inwhich order, simply click and holdeach 'reading' circle in turn, dragonto the image and adjust the size
accordingly.
5
When complete click ‘Save’ andclick the 'x' button in the top right toclose the design window.

www.elcometer.com

en-12

R

15 APPENDIX 2: ELCOMASTER® COLLECT DESIGN (continued)
To use a ‘Collect Design’:
1
Click on the ‘View’ tab and select afolder or project.
2
Click on ‘New’à’Batch Item’à’Dry
Film Thickness’ and select therequired Collect Design. A new
batch will be created. Unlike
standard batches it will have anadditional tab with the same nameas the Collect Design, click on thisto show your design.
3
To begin collection, click on‘Connect Gauge’ in the ‘Live
Readings’ window.
4
Select a gauge from the list†.
5
Select the relevant connection
method.
ElcoMaster® will link the selectedgauge and, when ready, display thegauge information within the Live
Readings window.
Once connected, green arrows will
flash around the first ‘reading’ circle tomeasure. When a measurement istaken, the ‘reading’ circle will change toshow the measured value‡, and thegreen arrows will move to the next
reading.

†‡

The gauge must be connected to ElcoMaster® before it will appear in the list.
The ‘reading circle’ will be green if the measurement is within set limits or no limits have been set and red,
if outside set limits.

TMA-0610 Issue 09 - Text with cover 25915

R

Guía del usuario
Elcometer 311
Medidor de acabado de pintura
en automóvileswww.elcometer.com

R

CONTENIDO
Sección

Página1

Descripción general del medidor

es-2

2

Contenido de la caja

es-2

3

Utilización del medidor

es-3

4

Introducción

es-3

5

Toma de una lectura

es-5

6

Estadísticas - Modelo T

es-6

7

es-6

8

Calibración del medidor
Descarga de datos - Modelo T

9Actualización del medidor - Modelo T

es-8

10

Repuestos y accesorios

es-8

11

Declaración de garantía

es-8

12

Especificaciones técnicas

es-9

13

Avisos legales e información sobre la normativa

es-10

14

Apéndice 1: Resumen de iconos del medidor

es-11

15

Apéndice 2: ElcoMaster® Collect Design

es-12

Pb

Hg

es-7

Para despejar cualquier duda, consulte la versión original en inglés.
Dimensiones del medidor: 141 x 73 x 37mm (5,55 x 2,87 x 1,46 pulgadas)
Peso del medidor:
156g (5,5 onzas) incluidas pilas
Patentes aplicables:
Medidores FNF: US5886522; Medidores Modelo T: US7606671© Elcometer Limited 2015 - 2024. Todos los derechos reservados. Este documento ni ningún fragmento del mismo
pueden reproducirse, transmitirse, transcribirse, almacenarse (en un sistema de recuperación o de otro tipo) ni
traducirse a ningún idioma, en ningún formato ni por ningún medio (ya sea electrónico, mecánico, magnético,óptico, manual o de otro tipo) sin permiso previo y por escrito de Elcometer Limited.

es-1

www.elcometer.com

R

1 DESCRIPCIÓN GENERAL DEL MEDIDOR
6

1
2
7
3
8

4

5

1
2
3
4
5
6
7
8

Indicadores LED: Rojo (izquierdo), Verde (derecho)
Pantalla Color
Teclas multifunciones
Tecla de encendido/apagado
Sonda integral
Conector de salida de datos USB (bajo la cubierta)
Compartimento de pilas (apertura/cierre mediante un ¼ de vuelta)
Conexión de correa para colgar de la muñeca

2 CONTENIDO DE LA CAJA
<
<
<
<
<
<
<
<

a

Medidor de acabado de pintura en automóviles Elcometer 311
Pieza de comprobación de calibración; acero y aluminio (solo medidores FNF)
Lámina de comprobación de calibración; 125 μm (5 mils)a
Arnés para muñeca
Cubierta protectora
2 pilas AA
Certificado de prueba
Guía del usuario

Valor nominal

www.elcometer.com

es-2

R

3 UTILIZACIÓN DEL MEDIDORa

Alimentación: Pilas o USB - incluidohaBluetooth activado (Modelo T) - Gris: nobemparejado, Naranja: emparejadocindicador de duración de pilasbcTipo de sustrato - F: Ferroso; FNF: Ferroso;
F N F: No ferrosodUnidades de medida - µm, milseTecla programada Menúf

Tecla programada Estadísticas(Modelo T) - consulte la página es-6
Tecla programada Calibración
Estadísticas

g
h

d

e
g

f

4 INTRODUCCIÓN
4.1 ASEGURANDO QUE SU MEDIDOR DISPONGA DELULTIMO FIRMWARE Y ACTUALIZANDO SU MEDIDOR MODELO T
Para asegurar que su medidor disponga del firmware más
actualizado, permitiéndole beneficiarse de las ultimas características yfuncionalidad, le recomendamos conectar regularmente el equipo a
ElcoMaster® y antes de su primer uso.
Simplemente conecte el medidor vía USB a un ordenador conectado
a internet y ejecutando ElcoMaster® utilizando la función ‘Conectar
medidor’. Si hay disponible una versión más actualizada del firmware
del medidor, la función ‘Actualizar medidor’ se mostrará a la derechade los detalles del medidor. Haga clic en ‘Actualizar medidor’ para
instalar el último firmware.
4.2 INSTALACIÓN DE LAS PILAS
Cada medidor se suministra con 2 pilas AA
alcalinas.
Para introducir o sustituir las pilas:
1
Levante el pestillo de la tapa delcompartimento de las pilas y gírelo en
sentido antihorario para retirar la tapa.
2Introduzca 2 pilas asegurándose de que lapolaridad sea correcta.
3
Vuelva a colocar la tapa y gire el pestillo
en sentido horario para cerrarlo.
es-3

+

+

www.elcometer.com

R

4 INTRODUCCIÓN (continuación)
El estado de las pilas se indica mediante un símbolo situado en laparte superior derecha de la pantalla ():
4 Símbolo de pila llena (naranja) = pilas totalmente cargadas
4 Símbolo de pila vacía (rojo intermitente) = pilas con el mínimo

nivel sostenible
4.3 ENCENDIDO/APAGADO
Para encender: Mantenga pulsado el botón de encendido/apagado
hasta que aparezca el logotipo de Elcometer.
Para apagar: Mantenga pulsado el botón de encendido/apagado
hasta que la pantalla quede en blanco.
Nota: El medidor se apagará automáticamente después de 1 minuto de inactividad.

4.4 AJUSTE DEL VOLUMEN DEL PITIDOEl medidor pita cada vez que se toma una medida. Para ajustar elvolumen del pitido:
1
2
3

Pulse la tecla programada ‘ ’.
Utilice las teclas programadas éê para resaltar ‘ ’ y pulse ‘ ’para seleccionar.
Utilice las teclas programadas éê para ajustar el volumen ypulse ‘ ’ para establecer o ‘ ’ para ignorar los cambios
realizados y regresar a la pantalla anterior.

4.5 AJUSTE DEL BRILLO DE LA PANTALLAEl brillo de la pantalla puede ajustarse como ‘ ’ (Automático) -el brillo
se ajusta automáticamente utilizando el sensor de luz ambiental delmedidor- o puede ajustarlo el usuario en función de las condiciones demedición. Para ajustar el brillo de la pantalla:
1
2
3

Pulse la tecla programada ‘ ’.
Utilice las teclas programadas éê para resaltar ‘ ’ y pulse ‘ ’para seleccionar.
Utilice las teclas programadas éê para ajustar el brillo de lapantalla y pulse ‘ ’ para establecer o ‘ ’ para ignorar loscambios realizados y regresar a la pantalla anterior.
4 El ajuste predeterminado es ‘ ’ (Automático).

www.elcometer.com

es-4

R

4 INTRODUCCIÓN (continuación)
4.6 SELECCIÓN DE LA UNIDAD DE MEDIDAEl Elcometer 311 puede realizar mediciones en μm y mils. Para
ajustar la unidad de medida:
1
2

Pulse la tecla programada ‘ ’.
Utilice las teclas programadas éê para resaltar ‘μm/mil’ y pulse‘ ’ para alternar entre ‘μm’ y ‘mil’, seguido de la tecla
programada ç para regresar a la pantalla de lectura principal.

5 TOMA DE UNA LECTURA
Antes de realizar una medición, es conveniente comprobar el
funcionamiento del medidor utilizando el procedimiento de comprobaciónde la calibración; consulte la sección 7.2, ‘Comprobación de la calibracióndel medidor’, en la página es-6.
Para tomar una lectura:
1
Mientras sujeta el medidor perpendicularmente, sitúe suavemente
la sonda sobre la superficie.
4 La lectura podría no ser correcta si la sonda no queda plana contra la

superficie.

2

En la pantalla se mostrará el espesor del revestimiento.
4 ‘>500µm’ (‘>20mils’) indica una lectura situada fuera del rango delmedidor.

3

Para lecturas posteriores, retire la sonda y vuelva a colocarla sobrela superficie revestida.

Empleando ElcoMaster®, el Elcometer 311 Modelo T puede transmitir
simultáneamente la lectura en vivo a un PC o dispositivo móvil para
generar un informe; consulte la sección 8, ‘Descarga de datos’, en lapágina es-7 para obtener más información.
El Elcometer 311 Modelo T calcula y muestra valores estadísticos altomar cada lectura; consulte la sección 6, ‘Estadísticas’, en la página
es-6 para más detalles.

es-5

www.elcometer.com

R

6 ESTADÍSTICAS - MODELO TEl Elcometer 311 Modelo T calcula y muestra los siguientes valores
estadísticos conforme se toma cada medida:ŸŸŸŸNúmero de lecturas (n:)
Lectura media (X:)
Lectura más baja (Lo:)
Valor de índice de Elcometer (EIV:)
Se utiliza para evaluar la calidad global de un revestimiento. Idóneo para su uso en el
sector de acabado de automóviles. Número de patente US7606671.

Para mostrar u ocultar las estadísticas de la pantalla, pulse la tecla
programada ‘’.
Para borrar los valores estadísticos actuales, mantenga pulsada la tecla
programada ‘’ durante aproximadamente 3 segundos (cuando semuestren estadísticas). El medidor pitará y las estadísticas se pondrán a
cero.
7 CALIBRACIÓN DEL MEDIDOR
7.1 UTILIZACIÓN DE LA CALIBRACIÓN CEROEl usuario puede clalibrar el medidor usando la calibración cero, unmétodo muy sencillo para calibrar en superficies rasas.
1
2

Pulse la tecla programada ‘ ’.
Cuando se le indique, mientras sujeta el medidor
perpendicularmente, sitúe suavemente la sonda sobre el sustrato
metálico no revestido. Aparecerá ‘Zero’ y el medidor regresará a lapantalla de lectura principal.

Nota: Los medidores FNF deben calibrarse tanto en modo ferroso como en modo no
ferroso para garantizar la precisión de las lecturas.

7.2 COMPROBACIÓN DE LA CALIBRACIÓN DEL MEDIDOREl usuario puede comprobar la calibración y el funcionamiento delmedidor empleando la lámina y la(s) pieza(s) de comprobación de lacalibración suministradas.
1
23

Coloque la lámina de comprobación de la calibración sobre la piezade comprobación.
Mientras sujeta el medidor perpendicularmente, sitúe suavemente
la sonda sobre la lámina.
El medidor debería leer el valor impreso en la etiqueta de la lámina,±25 μm (1 mil).

Si el medidor no muestra una lectura válida, puede que sea necesario
recalibrar; consulte la sección 7.1.
es-6
www.elcometer.com

R

8 DESCARGA DE DATOS - MODELO T
8.1 UTILIZACIÓN DE ELCOMASTER® EN UN PC
Al utilizar ElcoMaster® -disponible como descarga gratuita enelcometer.com- el Elcometer 311 Modelo T puede transmitir
simultáneamente la lectura dinámicas – en vivo a través de USB oBluetooth® a un PC para generar un informe. Para obtener más
información sobre ElcoMaster®, visite www.elcometer.com
8.2 CON APLICACIONES MÓVILES ELCOMASTER®
Las aplicaciones móviles ElcoMaster®, idóneas para su uso en campoo en las instalaciones, permiten a los usuarios:ŸŸ

Almacenar lecturas dinámicas directamente en un dispositivo
móvil y guardarlas en lotes junto con coordenadas de GPS.
Añadir fotografías de la superficie sometida a prueba (a través dela cámara del dispositivo móvil).

Ÿ

Mapear las lecturas sobre una fotografía o diagrama (al utilizar la®

función ‘Collect Design’ en ElcoMaster - consulte el Apéndice 2,‘ElcoMaster® Collect Design’, en la página es-12)
Ÿ

Los datos de inspección pueden transferirse de móvil a PC pararealizar análisis adicionales y generar informes.

Para obtener más información sobre las aplicaciones móviles
ElcoMaster®, visite www.elcometer.com

es-7

www.elcometer.com

R

9 ACTUALIZACIÓN DEL MEDIDOR - MODELO T
El usuario puede actualizar mediante ElcoMaster® el firmware a laversión más reciente cuando esté disponible. ElcoMaster® informará alusuario de la existencia de cualquier actualización cuando el medidor
se conecte a un PC con conexión a Internet.
Para actualizar el firmware del Elcometer 311 Modelo B, póngase en
contacto con su distribuidor o centro de servicio Elcometer más
cercano.
10 REPUESTOS Y ACCESORIOS
Los siguientes accesorios y repuestos se encuentran disponibles a
través de su distribuidor Elcometer local o directamente de Elcometer.
Descripción
Pieza de comprobación de calibración;
acero (ferroso)
Pieza de comprobación de calibración;
aluminio (no ferroso)
Lámina de comprobación de calibración;
125 μm (5 mils)a
Cable USB
Protectores de pantalla autoadhesivos, lote de 10

Número de pieza
T99916925
T99916901
T99022570-7A
T99921325
T99922341

11 DECLARACIÓN DE GARANTÍA
Los medidores Elcometer 311 se suministran con una garantía de un
año para defectos de fabricación que excluye contaminación ydesgaste.
La garantía puede ampliarse hasta dos años en un plazo de 60 días
después de la compra a través de www.elcometer.com.

aValor nominal

www.elcometer.com

es-8

R

12 ESPECIFICACIONES TÉCNICAS
Rango

0 - 500µm (0 - 20mils)

Precisiónb±5% o ±20µm (1,0mil)

Resolución

10µm (0,5mil)

Espesor mínimo delsustratoc

Acero:
Aluminio:

800µm (30mils)
300µm (12mils) - (solo medidores FNF)

Velocidad demedición

Hasta 60 lecturas por minuto

Temperaturas detrabajoDe -10 a 50°C (De 14 a 122°F)

Humedad relativa

De 0 a 95%

Fuente dealimentación(también pueden utilizarse pilas recargables)

Duración de laspilasdAlcalinas: Aproximadamente 16 horas
Litio: Aproximadamente 24 horas

Dimensiones delmedidor

14,1 x 7,30 x 3,70cm(5,55 x 2,87 x 1,46 pulgadas)

Peso del medidor

156g (5,5 onzas)(incluidas pilas)

2 pilas AA o a través de cable USB

Cumple las siguientes normas:
ISO 2178, ISO 2808, ISO 2808-7C, ISO 2808-7D, ISO 2808-12A,
ISO 2808-11B, ASTM E376, JIS K 5600-1-7, AS/NZS 1580.108.1b

El que sea mayor.
Para la precisión especificada.
dPuede diferir con pilas recargables.
ces-9

www.elcometer.com

R

13 AVISOS LEGALES E INFORMACIÓN SOBRE LA NORMATIVA
Declaración de conformidad:
Elcometer 311 Modelo B cumple los requisitos de las siguientes Directivas de la UE:
2014/30/EU
Directiva de compatibilidad electromagnética
2011/65/EU
Directiva Sobre Restricciones del Uso de Determinadas Sustancias Peligrosas en
Equipos Eléctricos y Electrónicos (RoHS)
Elcometer 311 Modelo T cumple los requisitos de las siguientes Directivas de la UE:
2014/53/EU
Radyo Ekipmanları
2014/30/EU
Elektromanyetik uyumluluk
2011/65/EU
Bazı tehlikeli maddelerin kullanımının sınırlandırılması
Las Declaraciones de Conformidad pueden descargarse a través de:Modelo B: https://downloads.elcometer.com/Declaration_of_Conformity/Spanish/DoC_311C_B.pdf
Modelo T: https://downloads.elcometer.com/Declaration_of_Conformity/Spanish/DoC_311C_T.pdf
Banda de frecuencias de funcionamiento: 2,402 - 2,480 MHz
Potencia máxima transmitida: <4 dBm
Este producto es un equipo de Clase B, Grupo 1 ISM, conforme a las normas CISPR 11.
Producto de clase B: Es apto para su uso en entornos domésticos y establecimientos conectados directamente a una red desuministro de baja tensión que suministre a edificios dedicados a uso residencial.
Producto de Grupo 1 ISM: Producto que genera y/o utiliza intencionadamente energía de radiofrecuencia de acoplamiento
conductivo necesaria para el funcionamiento interno del propio equipo.
El USB es para transferencia de datos solamente y no debe conectarse a la red eléctrica mediante un adaptador de USB/redeléctrica.
Este dispositivo cumple los requisitos de la parte 15 de las normas de la FCC. Su utilización está sujeta a las siguientes dos
condiciones: (1) Este equipo no puede provocar interferencias nocivas, y (2) este equipo debe aceptar cualquier interferencia recibida,incluidas las interferencias que puedan provocar un funcionamiento no deseado.
Elcometer 311 Modelo T: Puede accederse a la ID de FCC y la QDID de Bluetooth SIG pulsando ‘ ’ seguida de ‘ ’.
Elcometer 311 Modelo B: Se puede acceder a la información sobre la normativa manteniendo pulsado el botón de
ENCENDIDO/APAGADO mientras el medidor está apagado hasta que se muestre el menú, seguido de ‘ ’.
NOTA: Este equipo ha sido sometido a pruebas que confirman su cumplimiento de los límites para dispositivos digitales declase B, conforme a la parte 15 de las normas de la FCC. Estos límites están diseñados para proporcionar una protección
razonable frente a interferencias dañinas en instalaciones domésticas. Este equipo genera, utiliza y puede irradiar energía deradiofrecuencia, por lo que, si no se instala y utiliza conforme a las instrucciones, puede provocar interferencias dañinas en
comunicaciones de radio. No obstante, no existe garantía alguna de que no se produzcan interferencias en instalaciones
concretas. En el caso de que este equipo provoque interferencias dañinas en la recepción de radio o televisión, lo que puede
determinarse encendiendo y apagando el equipo, el usuario deberá intentar corregir dichas interferencias adoptando una ovarias de las siguientes medidas:-- Reoriente o cambie de lugar la antena receptora.
-- Aumente la distancia entre el equipo y el receptor.
-- Conecte el equipo a una toma de un circuito distinto al del receptor.
-- Consulte a su proveedor o a un técnico con experiencia en radio / TV para obtener ayuda.
Para cumplir los requisitos de exposición a radiofrecuencias de la FCC para dispositivos móviles y de transmisión de estaciónbase, debe mantenerse una separación mínima de 20 cm entre la antena de este dispositivo y las personas durante su
utilización. Para garantizar dicho cumplimiento, no se recomienda su utilización a una distancia inferior a ésta. La(s) antena(s)utilizada(s) para este transmisor no debe(n) situarse ni utilizarse junto a otra antena o transmisor.
Las modificaciones realizadas sin aprobación expresa de Elcometer Limited podrían anular la autorización concedida al
usuario para utilizar el equipo conforme a las normas de la FCC.
Este dispositivo cumple la(s) norma(s) RSS de exención de licencia de Industry Canada. Su utilización está sujeta a lassiguientes dos condiciones: (1) Este equipo no puede provocar interferencias, y (2) este equipo debe aceptar cualquierinterferencia recibida, incluidas las interferencias que puedan provocar un funcionamiento no deseado del dispositivo.
Este aparato digital de Clase B cumple la norma CAN ICES-3 (B)/NMB-3(B).
y ElcoMaster® son marcas comerciales registradas de Elcometer Limited, Edge Lane,
Mánchester, M43 6BU. Reino Unido.
son marcas comerciales propiedad de Bluetooth SIG Inc para las que se ha concedido
licencia a Elcometer Limited.
Todas las demás marcas comerciales se dan por reconocidas.
R

Sede central: Elcometer Limited, Edge Lane, Manchester, M43 6BU. Reino Unido

www.elcometer.com

es-10

R

14 APÉNDICE 1: RESUMEN DE ICONOS DEL MEDIDOR
Icono

Descripción

ModeloAlimentación: Pilas - incluido indicador de duración de pilas B, T
Alimentación: USB

B, T

Bluetooth activado - Gris: no emparejado, Naranja: emparejado TF,
Tipo de sustrato - F; ferroso (Modelo B), Medidores FNF
F NF, F N F (detección automática) - FNF: Ferroso; FNF: No ferroso

B, T

µm

Unidades de medida - µm

B, T

mils

Unidades de medida - mils

B, T

Tecla programada MenúB, T

Tecla programada Estadísticas

T

Tecla programada Calibración

B, T

Opción de menú Unidades de medida

B, T

Opción de menú Ajuste de volumen

B, T

Opción de menú Brillo de pantalla

B, T

Opción de menú Información legal

T

Tecla programada Seleccionar / Establecer

B, T

Tecla programada Escape

B, T

Opción de menú Datos de contacto de Elcometer T

es-11

www.elcometer.com

R

15 APÉNDICE 2: ELCOMASTER® COLLECT DESIGN
La función ‘Collect Design’ (diseño de®
recogida) de ElcoMaster permite a losusuarios establecer una plantilla de recogidapara indicar a los inspectores dónde y en qué
orden deben tomarse las lecturas.
Esta imagen puede utilizarse como guía derecogida en ElcoMaster® en un PC o puedetransferirse a un dispositivo móvil®
empleando la aplicación móvil ElcoMaster .
Al crear un nuevo ‘Collect Design’ (diseño de recogida), el usuario:
Ÿ
Ÿ
Ÿ
Ÿ
Ÿ

Añade la imagen requerida para la que se desean recoger lecturas;
Especifica el número de lecturas que deben tomarse;
Introduce cualquier límite que sea pertinente;
Puede solicitar un mensaje para comenzar el siguiente juego delecturas o que este comience automáticamente;
Elige entre crear un nuevo lote para cada juego de lecturas o crear un
solo lote con múltiples lecturas.

Para establecer un nuevo ‘Collect Design’:
1
Cambie ElcoMaster® al modoavanzado (‘Advanced Mode’)
haciendo clic en la flecha deampliación y seleccionando‘Advanced Mode’.
2
Haga clic en ‘Library’(Biblioteca)à’New’ (Nueva)à’Collect
Design’ (Diseño de recogida).
3
Siga las indicaciones de la pantalla yhaga clic en ‘Finish’ (Finalizar) para
comenzar el diseño.
4
Para indicar en la imagen dónde debe
tomarse cada lectura y en qué orden,
simplemente haga clic y mantenga
pulsado en cada círculo de 'lectura' deuno en uno, arrastre hasta la imagen yajuste el tamaño según corresponda.
5
Cuando haya terminado, haga clic en‘Save’ (Guardar) y haga clic en elbotón 'x' de la parte superior derechapara cerrar la ventana de diseño.

www.elcometer.com

es-12

R

15 APPENDIX 2: ELCOMASTER® COLLECT DESIGN (cont.)
Para utilizar un ‘Collect Design’ (diseño
de recogida):
1
Haga clic en la ficha ‘View’ (Ver) y
seleccione una carpeta o proyecto.
2
Haga clic en ‘New’ (Nuevo)à’Batch
Item’ (Elemento de lote)à’Dry Film
Thickness’ (Espesor de película seca)
y seleccione el diseño de recogida(Collect Design) requerido. Se creará
un nuevo lote. A diferencia de los lotes
estándar, tendrá una ficha adicional
con el mismo nombre que el diseño derecogida (Collect Design); haga clic en
ella para mostrar su diseño.
3
Para comenzar la recogida, haga clicen ‘Connect Gauge’ (Conectar
medidor) en la ventana ‘Live
Readings’ (Lecturas en vivo).
4
Seleccione un medidor de la lista†.
5
Seleccione el método de conexión
correspondiente.
ElcoMaster® enlazará el medidor
seleccionado y, cuando esté listo, mostrarála información del medidor en la ventana
Live Readings (Lecturas en vivo).
Una vez conectado, parpadearán unas
flechas verdes alrededor del primer círculo
de ‘lectura’ a medir. Al realizar una
medición, el círculo de ‘lectura’ cambiará
para mostrar el valor medido‡ y las flechas
verdes se trasladarán a la siguiente lectura.

†‡

El medidor debe estar conectado a ElcoMaster® para que aparezca en la lista.
El ‘círculo de lectura’ será verde si la medición está dentro de los límites establecidos o si no se han
establecido límites, y rojo si está fuera de los límites establecidos.

TMA-0610-04 emisión 09 - Texto con Cubierta 25915

R

Kullanım Kılavuzu
Elcometer 311
Otomotiv Boya Ölçüm Cihazı

www.elcometer.com

RİÇİNDEKİLER
Bölüm

Sayfa

1

Ölçüm Cihazına Genel Bakış

tr-2

2

Kutunun İçindekiler

tr-2

3

Ölçüm Cihazını Kullanma

tr-3

4

Başlangıçtr-3

5Ölçüm Yapma

tr-5

6

İstatistik - T Modeli

tr-6

7

Cihaz Kalibrasyonu

tr-6

8

Veri İndirme - T Modeli

tr-7

9

Ölçüm Cihazınızı Yükseltme - T Modeli

tr-8

10

Yedek Parçalar ve Aksesuarlar

tr-8

11

Garanti Bildirimi

tr-8

12

Teknik Özellikler

tr-9

13

Yasal Uyarılar ve Mevzuat Bilgileri

tr-10

14

Ek 1: Ölçüm Cihazı Simgelerinin Özeti

tr-11

15

Ek 2: ElcoMaster® Collect Design

tr-12

Pb

Hg

Şüpheye düştüğünüz durumlarda, lütfen orijinal İngilizce dil sürümüne bakın.
Ölçüm Cihazı Boyutları: 141 x 73 x 37mm (5,55 x 2,87 x 1,46 inç)Ölçüm Cihazı Ağırlığı:156kg (5,5 ons) piller dahil
Geçerli Patentler:
FNF Ölçüm Cihazları: US5886522; T Modeli Ölçüm Cihazları: US7606671© Elcometer Limited 2015 - 2024. Tüm hakları saklıdır. Elcometer Limited'in önceden yazılı izni olmaksızın, bu
belgenin hiçbir bölümü herhangi bir biçimde veya (elektronik, mekanik, manyetik, optik, manuel veya başka türlü)herhangi bir yolla çoğaltılamaz, aktarılamaz, kopyalanamaz, (bir geri alma sisteminde veya başka suretle)saklanamaz ya da herhangi bir dile tercüme edilemez.

tr-1

www.elcometer.com

R

1 ÖLÇÜM CIHAZINA GENEL BAKIŞ
6

1
2
7
3
8

4

5

1
2
3
4
5
6
7
8

LED Göstergeler - Kırmızı (sol), Yeşil (sağ)
Renkli EkranÇokişlevli Programlanabilir Tuşlar
Aç/Kapa TuşuBütünleşik Prob
USB Veri Çıktı Soketi (kapağın altında)Pil Bölümü (¼ döndürerek açma/kapama)
Bileklik Bağlantısı

2 KUTUNUN İÇINDEKILER<<

Elcometer 311 Otomotiv Boya Ölçüm Cihazı
Kalibrasyon Kontrol Parçası; Çelik ve/veya Alüminyum(yalnızca FNF cihazlarda)

<
<
<
<
<
<

a

Kalibrasyon Kontrol Folyosu; 125μm (5mil)a
Bileklik
Koruyucu Kılıf
2 x AA Pil
Test Sertifikası
Kullanım Kılavuzu

Nominal Değer

www.elcometer.com

tr-2

R

3 ÖLÇÜM CIHAZINI KULLANMAa

Güç: Pil veya USB -ahpil durumu göstergesi dahildirbcBluetooth Açık (T Modeli) -b

Gri: eşleştirilmiş; Turuncu: eşleştirilmemişc

Alt Taban Türü F: Demirli; F NF: Demirli; F N F: Demir dışıdÖlçü Birimleri - µm, milseMenü Tuşu

fİstatistik Tuşu (T Modeli) - tr-6. sayfaya
bakın

g
h

d

e
g

f

Kalibrasyon Tuşuİstatistik

4 BAŞLANGIÇ
4.1 CİHAZINIZIN SON SÜRÜMDE OLDUĞUNU SAĞLAMAK VECİHAZINIZI MODEL-T YE YÜKSELTMEKCihazınızın güncel sürümde oluğunu sağlama ve güncel sürümün
getirebileceği faydalardan yararlanmak için, cihazınızı ilk kullanımda
ve daha sonra belirli aralıklarda ElcoMaster® programına bağlamanızı
tavsiye ederiz.
Basitçe, internete bağlı bilgisayarınızda kurulu olan ElcoMaster®
yazılımında ‘Connect Gauge’ adımını kullanarak USB ile bilgisayarabağlayınız. Eğer, güncel sürüm var ise ‘Update Gauge’ yazısı cihazbilgilerini içeren bölümün sağında belirecektir. Güncel sürümü
yüklemek için ‘Update Gauge’ tıklamanız yeterli olacaktır.
4.2 PİLLERİ YERLEŞTİRME
Her cihaz 2 x AA alkalin pille birlikte gelir.
+

Pilleri yerleştirmek veya değiştirmek için:
1
Pil bölümünün kapağindaki mandalı
kaldırın ve kapağı çıkarmak için saatin
aksi yönünde çevirin.
2
Kutupların doğru oturmasına dikkat
ederek 2 pili yerleştirin.
3Kapağı yerine yerlşetirin ve kapatmak
için mandalı saat yönünde çevirin.
tr-3+

www.elcometer.com

R

4 BAŞLANGIÇ (devam)
Pilin durumu ekranın sağ üst köşesindeki bir simgeyle (gösterilir:)

4 Dolu simge (turuncu) = piller tam dolu
4 Boş simge (yanıp sönen kırmızı) = piller en düşük güç

düzeyinde
4.3 AÇMA/KAPATMA
Açmak için: AÇ/KAPA düğmesine basıp Elcometer logosu
görüntülenene kadar basılı tutun.
Kapatmak için: AÇ/KAPA düğmesine basıp ekran kapanana kadar
basılı tutun.
Not: Cihaz 1 dakika boş kalması durumunda otomatik olarak kapanacaktır.

4.4 BİP SESİ AYARIHer ölçüm yapıldığında cihaz bip sesi çıkarır. Bip sesinin düzeyini
ayarlamak için:
1
2
3‘ ’ tuşuna basın.
‘ ’ yi vurgulamak için éê tuşlarını kullanın ve seçmek için ‘ ’ye basın.
Ses düzeyini ayarlamak için éê tuşlarını kullanın ve ayarlarda
herhangi bir değişiklik yapmak veya yapılan herhangi birdeğişikliği yok saymak için ‘ ’ tuşuna basın ve bir önceki ekrana
dönün ‘ ’.

4.5 EKRAN PARLAKLIĞINI AYARLAMA
Ekran parlaklığı ‘ ’ (Otomatik) ayarına getirilebilir - cihazın ortam ışığı
sensörü kullanılarak parlaklık otomatik olarak ayarlanır - veya ölçüm
koşullarına uyum sağlayacak şekilde kullanıcı tarafından ayarlanabilir.
Ekran parlaklığını ayarlamak için:
1
2
3‘ ’ tuşuna basın.
‘ ’ yi vurgulamak için éê tuşlarını kullanın ve seçmek için ‘ ’ yebasın.
Parlaklık düzeyini ayarlamak için éê tuşlarını kullanın ve ayarlarda
herhangi bir değişiklik yapmak veya yapılan herhangi bir değişikliği
yok saymak için ‘ ’ tuşuna basın ve bir önceki ekrana dönün ‘ ’.
4 Varsayılan ayar ‘ ’ dir (Otomatik).

www.elcometer.com

tr-4

R

4 BAŞLANGIÇ (devam)
4.6 ÖLÇÜ BİRİMLERİNİ SEÇME
Elcometer 311, μm ve mil biriminden ölçüm yapabilir. Ölçü birimlerini
ayarlamak için:
1
2‘ ’ tuşuna basın.
‘μm/mil’i vurgulamak için éê tuşlarını kullanın ve ‘μm' ile 'mil’arasında gidip gelmek için ‘ ’ ye basıp, daha sonra ana okuma
ekranına dönmek için ç tuşuna basın.

5 ÖLÇÜM YAPMADeğeri okumadan önce, kalibrasyon kontrol usulünü kullanarak cihazın
performansını kontrol etmek iyi bir uygulamadır; tr-6. sayfadaki 'ÖlçümCihazının Kalibrasyonunu Kontrol Etme' başlıklı 7.2. Kısıma bakın.
Ölçüm yapmak için:
1
Cihazı dikey tutarak probu nazikçe yüzeye yerleştirin.
4 Prob yüzeyde dümdüz duracak şekilde tutulmuyorsa değer yanlış

ölçülebilir.

2

Kaplama kalınlığı ekranda gösterilir.
4 ‘>500µm’ (‘>20mils’) cihaz ölçüm aralığının dışında kalan bir değeri

gösterir.

3

Takip eden okumalar için, probu kaldırıp kaplamalı alana tekrar
yerleştirin.

ElcoMaster® kullanan Elcometer 311 T Modeli rapor oluşturmak için canlıölçümü aynı anda PC'ye veya mobil cihaza aktarabilir; daha fazla bilgi
için tr-7. sayfadaki 'Veri İndirme' başlıklı 8. Kısıma bakın.
Elcometer 311 T Modeli her okuma sırasında istatistiksel değerleri
hesaplar ve görüntüler; ayrıntılı bilgi için tr-6. sayfadaki 'İstatistik' başlıklı
6. Kısıma bakın.

tr-5

www.elcometer.com

R

6 İSTATISTIK - T MODELIElcometer 311 T Modeli her ölçüm sırasında aşağıdaki istatistikseldeğerleri hesaplar ve görüntüler:ŸŸŸŸÖlçüm sayısı (n:)Ölçüm ortalaması (X:)
En düşük ölçüm (Lo:)
Elcometer İndeks Değeri (EIV:)
Kaplamanın genel kalitesini değerlendirmek için kullanılır. Otomotiv boya ölçüm
sektöründe kullanım için idealdir. Patent numarası US7606671'dir.

İstatistikleri ekranda görüntülemek veya gizlemek için, ‘basın.

’ tuşunaGeçerli istatistiksel değerleri silmek için, ‘’ tuşuna basıp yaklaşık 3
saniye boyunca basılı tutun (istatistikler görünene kadar). Cihaz
bipleyecek ve istatistikler sıfırlanacaktır.
7 CIHAZ KALIBRASYONU
7.1 SIFIR KALIBRASYONUNU KULLANARAK
Kullanıcı, sıfır kalibrasyon kullanarak cihazı kalibre edebilir, pürüzsüz
yüzeylerde basit kalibrasyon yöntemidir.
1
2‘ ’ tuşuna basın.
İstendiğinde, cihazı dik tutarak hafifçe probu kaplanmamış metal
yüzeye yerleştirin. ‘Sıfır’ görüntülenir ve cihaz ana okuma
ekranına geri döner.

Not: Ölçümlerin doğruluğunu sağlamak için FNF cihazlarda hem demir (F) hem
de demirdışı (FNF)modda kalibre edilmelidir.

7.2 ÖLÇÜM CIHAZININ KALIBRASYONUNU KONTROL ETME
Kullanıcı cihazla birlikte verilen kalibrasyon kontrol folyosunu veparçasını (parçalarını) kullanarak cihazın kalibrasyonunu ve
performansını kontrol edebilir.
12
3

Kalibrasyon kontrol folyosunu kontrol parçasının üzerine yerleştirin.
Cihazı dikey tutarak probu nazikçe folyonun üzerine yerleştirin.
Cihaz folyo etiketinin üzerindeki basılı değeri ±25μm (1mil)okuyacaktır.

Cihaz geçerli bir okuma görüntülemezse, tekrar kalibrasyon
gerektirebilir - Bölüm 7.1 bakınız.
www.elcometer.com

tr-6

R

8 VERI İNDIRME - T MODELI
8.1 ELCOMASTER®'I PC'DE KULLANMA
ElcoMaster®'ı kullanma - elcometer.com adresinden ücretsiz olarak
indirilebilir - Elcometer 311 T Modeli rapor oluşturma için canlı okumayı
aynı anda USB veya Bluetooth® üzerinden bir PC'ye aktarabilir.
ElcoMaster® hakkında daha fazla bilgi için www.elcometer.com
adresini ziyaret edin.
8.2 ELCOMASTER® MOBİL UYGULAMALARI KULLANMA
Sahada veya yerinde kullanmak için ideal olan ElcoMaster® Mobil
Uygulama ile kullanıcılar aşağıdakileri yapabilir:ŸŸŸ

Canlı ölçümleri doğrudan mobil cihazlara depolayabilir ve GPS
koordinatlarıyla birlikte bunları gruplar halinde saklayabilir.
Test yüzeyinin fotoğraflarını ekleyebilir (mobil cihazın kamerası ile).
Ölçümleri bir fotoğraf veya diyagram üzerinde haritalandırabilir(ElcoMaster®'ın 'Collect Design' işlevini kullanarak - tr-12. sayfadaki®
'ElcoMaster Collect Design' başlıklı Ek 2'ye bakın.)ŸÖlçüm verileri analiz ve raporlama için mobil cihazdan PC'ye
aktarılabilir.

ElcoMaster® Mobil Uygulamalar hakkında daha fazla bilgi içinaşağıdaki adresi ziyaret edin www.elcometer.com

tr-7

www.elcometer.com

R

12 TEKNIK ÖZELLIKLERÖlçüm Aralığı

0 - 500µm (0 - 20mils)

Hassasiyetb±%5 veya ±20μm (1,0mil)Çözünürlük

10µm (0,5mil)

Minimum Alt TabancÇelik:
Alüminyum:

800µm (30mils)
300µm (12mils) - (yalnızca FNF cihazlarda)Ölçüm Hızı

Dakikada 60 okumaya kadarÇalışma Sıcaklığı

-10 ila 50°C (14 ila 122°F)Bağıl Nem

0 ila %95

Güç Kaynağı

2 x AA Pil veya USB Kablo ile(şarj edilebilir pil de kullanılabilir)

Pil Ömrüd

Alkalin: Yaklaşık 16 saat
Lityum: Yaklaşık 24 saatÖlçüm Cihazı
Boyutları

141 x 73 x 37mm
(5,55 x 2,87 x 1,46 inç)Ölçüm Cihazı Ağırlığı
(piller dahil)

156kg (5,5 ons)Aşağıdakilere uygun olarak kullanılabilir:
ISO 2178, ISO 2808, ISO 2808-7C, ISO 2808-7D, ISO 2808-12A,
ISO 2808-11B, ASTM E376, JIS K 5600-1-7, AS/NZS 1580.108.1

b

Hangisi daha büyükse.
Belirlenmiş hassaslık düzeyi için.
dŞarj edilebilir piller farklılık gösterebilir.
ctr-9

www.elcometer.com

R

9 ÖLÇÜM CIHAZINIZI YÜKSELTME - T MODELICihazın donanım yazılımı kullanıcı tarafından ElcoMaster® üzerinden
piyasada bulunan en son sürümüne yükseltilebilir. Cihaz internetbağlantısı olan bir PC'ye bağlı olduğunda ElcoMaster® kullanıcıyı olası
güncellemelerden haberdar edecektir.
Elcometer 311 B modeli donanım yazılımını yükseltmek için, lütfen en
yakın Elcometer Dağıtımcınızla veya Servis Merkeziyle iletişime geçin.
10 YEDEK PARÇALAR VE AKSESUARLAR
Yerel Elcometer tedarikçinizden veya doğrudan Elcometer'danedinebileceğiniz yedek parça ve aksesuarlar aşağıdaki gibidir.
Tanım
Parça Numarası
Kalibrasyon Kontrol Parçası; Çelik (Demirli)
T99916925
Kalibrasyon Kontrol Parçası; Alüminyum (Demir dışı) T99916901
Kalibrasyon Kontrol Levhası; 125μm (5mil)a
T99022570-7A
USB Kablosu
T99921325
Kendinden Yapışkanlı Ekran Koruyucular, 10'lu Paket
T99922341
11 GARANTI BILDIRIMI
Elcometer 311 ölçüm cihazları imalat hatalarına karşı iki yıl garantili
sunulmaktadır; kirlenme, ve kullanıcıdan kaynaklı hatalar yıpranma
garantiye dahil değildir.

aNominal Değer

www.elcometer.com

tr-8

R

13 YASAL UYARILAR VE MEVZUAT BILGILERI
Uygunluk Beyanı:
Elcometer 311 Model B, aşağıdaki AB Direktiflerinin gerekliliklerine uygundur:
2014/30/EU
Elektromanyetik uyumluluk
2011/65/EU
Bazı tehlikeli maddelerin kullanımının sınırlandırılması
Elcometer 311 Model T, aşağıdaki AB Direktiflerinin gerekliliklerine uygundur:
2014/53/EU
Radyo Ekipmanları
2014/30/EU
Elektromanyetik uyumluluk
2011/65/EU
Bazı tehlikeli maddelerin kullanımının sınırlandırılması
Uygunluk Bildirimleri aracılığıyla şu adresten indirebilirsiniz:
Model B: https://downloads.elcometer.com/Declaration_of_Conformity/Turkish/DoC_311C_B.pdf
Model T: https://downloads.elcometer.com/Declaration_of_Conformity/Turkish/DoC_311C_T.pdf
Operasyonel Frekans Bandı: 2,402 - 2,480 MHz
Maksimum İletim Gücü: <4 dBm
Bu ürün, CISPR 11'e göre Sınıf B, Grup 1 ISM cihazıdır.
Sınıf B ürün: Ev içi işyerlerinde ve ev eşyaları için kullanılan binalarda kullanılan alçak gerilim güç şebekesine doğrudanbağlı işyerlerinde kullanım için uygundur.
Grup 1 ISM ürünü: Ekipmanın dahili çalışması için gerekli olan, bilerek üretilen ve / veya iletken olarak bağlanmış radyo
frekansı enerjisinin kullanıldığı ürün.
USB yalnızca veri aktarımı içindir ve bir USB şebeke adaptörü ile şehir şebekesine bağlanmamalıdır.
Bu cihaz, FCC Kurallarının 15. Bölümüne uygundur. Aşağıdaki iki koşula tabi olarak çalıştırılabilir: (1)bu cihaz zararlı girişimlere neden olmayacaktır ve (2) bu cihaz istenmeyen şekilde çalışmasına neden olabilecekgirişimler de dahil, alınan her türlü girişimi kabul edecektir.
Elcometer 311 T Modeli: FCC ID ve Bluetooth SIG QDID'ye önce ‘ ’ ardından ‘ ’ ye basılarak erişilebilir.
Elcometer 311 B Modeli: Açma/Kapama tuşuna ‘ ’ ve menu çıkana kadar basılarak ve basılı tutularak düzenleyici
bilgilere ulaşılabilir.
NOT: Bu ekipman test edilmiş ve FCC Kurallarının 15. Bölümü uyarınca B Sınıfı dijital cihaz limitlerine uygun olduğu
görülmüştür. Bu limitler yerleşim bölgesi kurulumlarında zararlı girişimlere karşı makul bir koruma sağlamak üzere
tasarlanmıştır. Bu ekipman radyofrekans enerjisi üretir, kullanır ve yayabilir ve talimatlara uygun şekilde kurulmaz ve
kullanılmazsa, radyo iletişimi için zararlı girişimlere neden olabilir. Bununla beraber, belli bir kurulumda girişim
oluşmayacağının garantisi yoktur. Bu ekipman radyo veya televizyon sinyali üzerinde zararlı girişimlere yol açarsa,
kullanıcının aşağıdaki bir veya daha fazla yöntemle girişimi gidermeyi denemesi önerilir:-- Alıcı antenin yönünü veya yerini değiştirin.
-- Ekipman ile alıcı arasındaki mesafeyi artırın.
-- Ekipmanı alıcının bağlı olduğundan farklı bir devredeki prize bağlayın.
-- Yardım için yetkili bayiye veya bir radyo/TV teknisyenine danışın.
Mobil ve baz istasyonu aktarım cihazlarına yönelik FCC RF Maruziyeti şartlarını yerine getirmek için, cihazın çalışması
esnasında cihazın anteni ile kişiler arasında en az 20 cm'lik bir mesafe bulunmalıdır. Şartlara uygunluk açısından,cihazın bu mesafeden daha yakında çalıştırılmaması önerilir. Bu verici için kullanılan anten(ler) başka bir anten veya
verici ile aynı yerde bulunmamalı veya birlikte çalıştırılmamalıdır.
Elcometer Limited tarafından özellikle onaylanmış olmayan değişiklikler, FCC kuralları uyarınca kullanıcının ekipmanıçalıştırma yetkisini hükümsüz kılabilir.
Bu cihaz Industry Canada lisans dışı RSS standardına (standartlarına) uygundur. Aşağıdaki iki koşula tabi olarakçalıştırılabilir: (1) bu cihaz zararlı girişimlere neden olmayacaktır ve (2) bu cihaz istenmeyen şekilde çalışmasına neden
olabilecek girişimler de dahil, alınan her türlü girişimi kabul edecektir.
Bu B Sınıfı dijital cihaz, CAN ICES-3 (B) / NMB-3 (B) ile uyumludur.
ve ElcoMaster®, Elcometer Limited'in tescilli ticari markalarıdır, Elcometer Limited, Edge Lane,
Manchester, M43 6BU, İngiltere
Bluetooth SIG'in sahip olduğu ve Elcometer Limited'in lisansını aldığı ticari markalardır.
Diğer tüm ticari markalar tanınmıştır.
R

Merkez: Elcometer Limited, Edge Lane, Manchester, M43 6BU, Birleşik Krallık.

www.elcometer.com

tr-10

R

14 EK 1: ÖLÇÜM CIHAZI SIMGELERININ ÖZETI
Simge

Tanım

ModeliGüç: Pil - Pil Ömrü Göstergesi dahilB, T

Güç: USB

B, T

Bluetooth Açık - Gri: eşleştirilmiş; Turuncu: eşleştirilmemişT

F,
Alt Taban Türü - F; Demirli (B Modeli)B, T
F NF, F N F FNF Cihazlar (Otomatik Algılama) - FNF: Demirli; FNF: Demir dışıµmÖlçü Birimleri - µm

B, T

mils

Ölçü Birimleri - mils

B, T

Menü Tuşu

B, Tİstatistik TuşuT

Kalibrasyon TuşuB, TÖlçü Birimleri Menü SeçeneğiB, T

Ses Ayarı Menü Seçeneği

B, T

Ekran Parlaklığı Menü Seçeneği

B, T

Yasal Bilgiler Menü Seçeneği

T

Seç / Ayarla TuşuB, TÇıkış TuşuB, T

Elcometer İletişim Bilgileri Menü Seçeneği

T

tr-11

www.elcometer.com

R

15 EK 2: ELCOMASTER® COLLECT DESIGN
ElcoMaster®'ın 'Collect Design' işlevi
kullanıcıların Denetçileri okumaların
nerede ve hangi sırayla alınacağı
konusunda yönlendirecek bir toplamaşablonu oluşturmalarına olanak verir.
Bu görüntü bir PC üzerinde
ElcoMaster®'da toplama kılavuzu olarak
kullanılabilir veya ElcoMaster® Mobil
Uygulama kullanılarak bir mobil cihazaaktarılabilir.
Kullanıcı, yeni bir 'Collect Design' oluştururken:
Ÿ
Ÿ
Ÿ
Ÿ
Ÿ

İlgili ölçümleri toplamak için gereken görüntüyü ekler;
Alınacak ölçüm sayısını belirler;İlgili limitleri girer;
Sıradaki ölçümleri başlatmak veya otomatik olarak başlatmak içinbilgi isteminde bulunabilir;
Her ölçüm kümesi için yeni bir grup oluşturmayı veya çok ölçümlü
tek bir grup oluşturmayı seçebilir.

Yeni bir 'Collect Design' oluşturmak için:
1
ElcoMaster®'ı 'Gelişmiş Mod' agetirmek için büyütme okuna
tıklayıp 'Gelişmiş Mod'u seçin.
2‘Kitaplık’à’Yeni’à’Collect Design’atıklayın.
3
Ekrandaki komutları izleyin ve
tasarımı başlatmak için 'Bitir'etıklayın.
4
Görüntüde her ölçümün nerede ve
hangi sırayla alınacağını göstermekiçin yalnızca sırasıyla her 'ölçüm'çemberine tıklayıp tutun ve
görüntünün üzerine sürükleyerek
boyutunu gerektiği gibi ayarlayın.
5
Bittiğinde 'Kaydet'e tıklayın vetasarım penceresini kapatmak içinsağ üst köşedeki 'x' düğmesinebasın.
www.elcometer.com

tr-12

R

15 EK 2: ELCOMASTER® COLLECT DESIGN (devam)
Bir 'Collect Design' kullanmak için:
1'Görüntüle' sekmesine tıklayın ve birklasör veya proje seçin.
2‘Yeni’à’Grup Unsuru’à’Kuru Film
Kalınlığı’na tıklayın ve istenen
Collect Design'ı seçin. Yeni bir grupoluşturulur. Standart gruplardan
farklı olarak, Collect Design ile aynı
ada sahip fazladan bir sekmesi
olacaktır, tasarımınızı görüntülemekiçin buna tıklayın.
3
Toplamayı başlatmak için, 'CanlıÖlçümler' penceresinde 'CihazıBağla'ya tıklayın.
4
Listeden bir cihaz seçin†.
5İlgili bağlantı yöntemini seçin.
ElcoMaster® seçilen cihazıbağlantılandırır ve hazır olduğunda, CanlıÖlçümler penceresinde cihaz bilgilerini
görüntüler.
Bağlandıktan sonra, ölçülecek ilk 'ölçüm'çemberinin etrafında yeşil oklar yanıp
söner. Bir ölçüm yapıldığında, 'ölçüm'çemberi ölçülen değeri‡ gösterecekşekilde değişecek, yeşil oklar da sıradakiölçüme geçecektir.

†‡

Cihazın listede görünmesi için ElcoMaster®'a bağlanmış olması gerekir.
Ölçüm belirlenen limitler dahilindeyse veya limit belirlenmemişse, 'ölçüm çemberi' yeşil, belirlenen
limitlerin dışındaysa, kırmızı olacaktır.

TMA-0610-14 Sayı 09 - Metin ve kapak 25915