50MHz Arbitrary Function Generator

FGX-295

FGX-295

50MHz Arbitrary Function Generator

50MHz Arbitrary Function Generator● Output range up to 50MHz for sine waveform and 25MHz forsquare waveform.
● 5 standard waveform: Sine, Square, Ramp, Pulse and Noise● Arbitrary waveform function with 14bits, 125Mega sampling
per sec.
● Flexible frequency sweep, burst waveform and modulation(AM, FM, PM, FSK and PWM) function.
● Standard equipment of USB, LAN and GP-IB● Standard equipment of a useful PC application● Memorizes up to 4kinds of an arbitrary waveform with 256
points.

Output waveform

TriggerStandardSine, Square, Ramp, Triangle, Pulse, Noise, DC

Incorporated arbitrary

Exponential rise, Exponential fall, Reverse ramp, Sinc, Heart Pulse

Trigger input

Waveform characteristics

Trigger output

Level

TTL compatible

Level

TTL compatible, Termination more than 1kΩ

Max. speed

1MHz

Sine

Frequency

1µHz to 50MHz

Parallel output

SquareFrequency

1µHz to 25MHz

Clock output

Frequency

1µHz to 50MHz

Ramp, Triangle

Frequency

1µHz to 200kHz

Data output

Level

TTL compatible, Termination more than 2kΩ

Pulse

Frequency

500µHz to 10MHz

General specification

Noise

Bandwidth

approx. 20MHz

Voltage range

100V to 240V ( 50Hz/60Hz ) 100V to 120V ( 400Hz )Frequency

1µHz to 10MHz

Power consumption

max. 80VA

Waveform length

2K to 256K points

Amplitude resolution

14bits ( including marks )Spec guaranty
temperature range

18 degree. To 28 degree.
( Humidity less than 80%, no condensation )Interface

USB ( Type-B ), GP-IB, LAN

Dimensions

224 (W)×107(H)×380(D)mm

WeightApprox. 3.6kg

Arbitrary

Sampling rate

Power source
Operational
circumstances

125MS/sOthersCommon characteristics
FrequencyResolution

1µHzAmplitude

Range

50Ω termination

10mV(p-p) to 10V(p-p)

DC offset, DC

Range (peak AC+DC)

50Ω termination±5V(p-p)Main output

Impedance

Approx. 50Ω

Sync output

Level

TTL ( Impedance : approx. 50Ω )Internal frequency
standard● Application Software "WAVEPATT"

Accuracy±10ppm ( 90days )This application software enables to create on the PC, an arbitrary

External frequency
standard (input)

Lock range

10MHz ±500Hz

Level

100mV(p-p) to 5V(p-p)

External frequency
standard (output)

Lock range

10MHz

Level

approx. 632mV(p-p) ( 0dBm )Range-360 degree to +360 degree

Resolution

0.001 degree

Internal modulation

Sine, Square, Ramp, Triangle, Noise, Arbitrary

Frequency(Internal)

2mHz to 20kHz

AM

Modulation

0.0% to 120.0%

FM

Deviation

DC to 25MHzPM

Deviation

0.0 degree to 360 degree

PWM

Deviation

0.0 degree to 360 degree

Internal modulation

duty cycle 50% square

Frequency(Internal)

2mHz to 100kHz

Voltage range±5V full scale

Input resistanceapprox. 8.7kΩ

Bandwidth

DC to 20kHz

Waveform

Sine, Square, Ramp, Arbitrary

Sweep time

1ms to 500 second

Phase offset±20ppm (1year )Modulation

FSK
External modulation
input
Sweep

Burst30

Limited sales area : China, Korea, Singapore,Malaysia, Thailand, Vietnam

Waveform

Sine, Square, Ramp, Triangle, Noise, Arbitrary

Start/Stop, Phase-360 degree to +360 degree

Internal period

1µs to 500 second

waveform that outputs from FGX-295. The created waveform is to betransmitted to FGX-295 through GP-IB, LAN or USB.