Elcometer 6910
elcometer.com

Flashpoint1

Flash Point
Setaflash ‘Series 3’ Closed Cup Tester

Elcometer 6910/1

The Elcometer 6910/01 Closed Cup Tester is an easy to use instrument that cancomplete a flash/no flash test in less than two minutes or determine the flash pointof a sample within a temperature range of ambient to 300°C (ambient to 572°F).
It provides fast and reliable flash point results and requires minimum operator skills.
Test parameters, instrument status and test results are accessed via the touch
screen display and a push button, which guide the operator through a straightforward
testing process.
An audible prompt is sounded when a flash is achieved. An automatic flash detector
reduces the chance of misinterpretation of the test result by the operator andenhanced repeatability. For easy record keeping the instrument stores 100,000 test
results, which can be saved and transferred via the USB port.
Gas for the test jet shutter assembly is supplied from an integral gas tank, which isfilled using a standard butane (lighter) refill cartridge.

STANDARDS:ASTM D1655, ASTM D3278,
ASTM D3828, ASTM E502,
BS 2000-523, CLP Regulations EC
No 1272/2008, DEF STAN 91-91,
EPA 1020 A & B, ISO 3679, ISO 3680,
UN Class 3 Non-viscous FlammableLiquids•
••••
•
••

Flash point test in less than 2 minutes
Small sample size, 2ml (0.067fl oz) or 4ml (0.135fl oz)
Automatic flash detection
Automatic barometric correction
USB port and result storage
Colour touch screen
Portable
Simple calibration

Technical Specification
Part Number

Description

K6910M010

Elcometer 6910/1 Setaflash ‘Series 3’ Closed Cup Tester

Temperature Range

Ambient to 300°C (Ambient to 572°C)

Sample Size

2ml (0.067fl oz) or 4ml (0.135fl oz)

Fuel Supply

Integrated gas tank (removable)

Cup MaterialAluminium

Dimensions

195 x 295 x 140mm (7.6 x 11.6 x 5.5”)

Data Storage and Download

1GB internal memory and integrated USB port

Pressure Correction

Automatic barometric correction

Weight3kg (6.6lb)

Packing List

Elcometer 6910 Setaflash ‘Series 3’ Closed Cup Tester, 2ml (0.067fl oz) syringe, gas valve &cap, viton ‘O’ ring, mains power cable, 2mm Allen key, quick start guide, USB memory stick withuser manual, quality assurance certificate and warranty registration formAccessories
KT006910N001

Cooling Module Option for 5°C (41°F) Ambient Temperature

KT006910N009

Certified Flash Point Material (50ml/1.7floz)○ Optional Calibration Certificate available.

2Certificate○Flash Point
Setaflash ‘Series 3’ Open Cup Tester

Elcometer 6910/2

The Elcometer 6910/2 Open Cup Tester offers the fastest and most accurate flash
point instrument at a cost effective price.
Certain substances, classified as “flammable” by closed cup flash point
testing, may be reclassified as “non-flammable” by combustibility testing. This has
significant potential cost reduction implications for the packaging, storage andshipping of many materials.
The Elcometer 6910/2 features an open cup for flash/no-flash finite
determinations or sustained combustion tests, audible and on-screen prompts andeasy calibration. Flash point tests can be conducted in less than two minutes witha temperature range from ambient to 300°C (ambient to 572°F).
A manually operated flame sweeping arm is fitted to the cup and gas is supplied
from the integral tank via a control valve. The flash or sustained combustion
characteristics of the sample are observed visually by passing the test flame overthe sample.
•••••

Flash point test in less than 2 minutes
Small sample size, 2ml (0.067fl oz)
Audible and on-screen prompts
Portable
Simple calibration

STANDARDS:ASTM D1655, ASTM D3278,
ASTM D3828, ASTM E502,
BS 2000-523, CLP Regulations EC No
1272/2008, DEF STAN 91-91,
EPA 1020 A & B, ISO 3679, ISO 3680,
UN Class 3 Non-viscous Flammable
Liquids

Technical Specification
Part NumberDescription

UK 240V

EUR 220V

US 110V

K0UK6910M011

K0006910M011

K0US6910M011

Temperature Range

Ambient to 300°C (Ambient to 572°F)

Sample Size

2ml (0.067fl oz)

Test Time

1 minute for flash points up to 100°C (212°F)

Elcometer 6910/2 Setaflash ‘Series 3’
Open Cup Tester

Certificate○2 minutes for flash points above 100°C (212°F)
Temperature
Measurement

0.5°C (1.0°F) resolution 0.5°C (1.0°F) accuracy

Cup Material

Blackened aluminium

Dimensions

260 x 280 x 260mm (10.2 x 11.0 x 10.2”)

Weight5kg (11.0lb)

Packing List

Elcometer 6910 Setaflash ‘Series 3’ Open Cup Tester, 2ml (0.067fl oz) syringe, gas valve & cap, 105mmsilicone tube, test flame height gauge, mains power cable, inspection certificate, warranty registration
form and operating instructions

Accessories
KT006910N001

Cooling Module Option for 5°C (41°F) Ambient Temperature

KT006910N009

Certified Flash Point Material (50ml/1.7fl oz)○ Optional Calibration Certificate available.

elcometer.com

3

Flash Point
Elcometer 6910/3

Setaflash ‘Series 3’ Active-Cool Closed Cup Tester
The Setaflash ‘Series 3’ Active-Cool Tester offers the same features as theElcometer 6910/1 Setaflash but is fitted with a cup suitable for testing corrosivesamples.
It has been designed to carry out flash/no flash tests rapidly and efficiently
to determine the flash point of liquids and semi-solids in the 10°C to 130°C(50°F to 266°F) temperature range. A sample cup is heated and then cooled by
Peltier cells. This gives very accurate and stable sub-ambient test temperatures andthe benefit of rapid temperature normalisation between tests.
Flash point is automatically detected using a thermally activated detector, reducing
the risk of operator error and minimising the potential danger of inhaling fumes
during a test. A rechargeable gas tank with on/off switch and fine adjustment areintegral to the unit.

STANDARDS:ASTM D1655, ASTM D3278,
ASTM D3828, ASTM D7236,
ASTM E502, BS 2000-523,
DEF STAN 91-91, EPA 1020 A & B,
ISO 3679, ISO 3680,
UN Class 3 Non-viscous FlammableLiquids•
••
•
••
•
•

Flash/No Flash & Ramp Modes
Flash point test in less than 2 minutes
Small sample size, 2ml (0.067fl oz) or 4ml (0.135fl oz)
Automatic flash detection
Electronic Peltier cooling
Audible and on-screen prompts
PortableCorrosion resistant cup insert

Technical Specification
Part Number

Description

K6910M013

Elcometer 6910/3 Setaflash ‘Series 3’ Active-Cool

Temperature Range

10°C to 135°C (50°F to 275°F)

Test Modes

Rapid Equilibrium and Ramp

Sample Size

2ml (0.067fl oz) for flash points up to 100°C (212°F)

Certificate○4ml (0.135fl oz) for flash points above 100°C (212°F)
Test Duration

1 minute below 100°C, 2 minutes above 100°C, or user determined 1 to 99 minutes

Ramp Rate2°C/min ramp (3.6°F/min ramp)

Cup Material

Corrosion resistant steel insert

Dimensions

195 x 295 x 140mm (7.6 x 11.6 x 5.5”)

Weight3kg (6.6lb)

Packing List

Elcometer 6910 Setaflash ‘Series 3’ Active-Cool Closed Cup Tester, 2ml (0.067fl oz) syringe,gas valve & cap, power supply, mains power cable, 2mm Allen key, quality assurance certificate,warranty registration form and operating instructions

Accessories
KT006910N009

Certified Flash Point Material (50ml/1.7fl oz)○ Optional Calibration Certificate available.

4elcometer.be • elcometer.fr • elcometer.deelcometer.nl • elcometer.jp • elcometer.com.sg© Elcometer Limited, 2019. All rights reserved. No part of this document may be reproduced, transmitted, stored (in a retrieval system or otherwise), or translated into any language, in any form, or by any means, without the prior written
permission of Elcometer Limited.

elcometer.com

5