enElcometer 1506-B

Cylindrical Mandrel Bend Tester

Operating Instructions

enR

is a registered trademark of Elcometer Limited
All other trademarks acknowledged.
© Copyright Elcometer Limited. 2023.
All rights reserved. No part of this Document may be reproduced, transmitted, transcribed, stored (in a retrieval
system or otherwise) or translated into any language, in any form or by any means (electronic, mechanical,magnetic, optical, manual or otherwise) without the prior written permission of Elcometer Limited.
A copy of this Instruction Manual is available for download on our Website via www.elcometer.com
Doc.No. TMA-0763 Issue 01
Text with Cover No:31978

is a registered trademark of Elcometer Instruments Ltd.
All other trademarks acknowledged.
© Copyright Elcometer Instruments Ltd. 2005.
All rights reserved. No part of this Document may be reproduced, transmitted, transcribed, stored(in a retrieval system or otherwise) or translated into any language, in any form or by any means

Section

Page

1

About your tester . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 2

2Testing a specimen. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4

3Technical specification. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 74Maintenance . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 7

5Accessories . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 86Related equipment . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 10

1en

RCONTENTS

en

R

T

hank you for your purchase of this Elcometer 1506-B Cylindrical Mandrel Bend Tester. Welcometo Elcometer.

Elcometer are world leaders in the design, manufacture and supply of inspection equipment for coatingsand concrete.

Our products cover all aspects of coating inspection, from development through application to post
application inspection.
The Elcometer 1506-B Cylindrical Mandrel Bend Tester, is a world beating product. With the purchase ofthis product you now have access to the worldwide service and support network of Elcometer. For
more information visit our website at www.elcometer.com

1 ABOUT YOUR TESTERThe Elcometer 1506-B Cylindrical Mandrel Bend Tester is a robust and simple-to-use instrumentfor determining the elasticity, adhesion and elongation of dried paint on sheet metal.
The metal sheet is clamped firmly into an adjustable vice and is bent perfectly over a mandrel by a set ofadjustable rollers. A range of mandrel sizes is available to allow the point at which coating failure occurs tobe determined. See “Accessories” on page 8.

2

The Elcometer 1506-B can be used in accordance with the following National and InternationalStandards:•

ASTM D 522-B supersedes ASTM D 1737•

AS/NZS 1580.402.1•

ISO 1519-2•

JIS K 5600-5-1

1.2 WHAT THE BOX CONTAINS• Elcometer 1506-B Cylindrical Mandrel Bend Tester•

Operating instructions

Note: Mandrels are not supplied as standard and must be ordered separately. See “Accessories” onpage 8.
To maximise the benefits of your new Elcometer 1506-B Cylindrical Mandrel Bend Tester pleasetake some time to read these Operating Instructions. Do not hesitate to contact Elcometer oryour Elcometer supplier if you have any questions.

3

en

R

1.1 Standards

enR

2 TESTING A SPECIMEN
1. Place the tester on a table with the black vice-tightening knob to your left.
2. Swing the bending lever anticlockwise until it reaches its stop position.
3. Select a mandrel and insert the mandrel into the recesses in the bending lever mechanism.
4. Swing the bending lever clockwise until it reaches its stop position.
231
5. Open the vice jaws and insert the specimen into the jaws of the vice. The painted surface of thespecimen should be facing away from the mandrel.

4

5

4

6
5

8

7

7. Slide the vice towards the mandrel until the specimen comes into contact with the mandrel.
8. Tighten the vice jaws until the specimen is held firmly.
If the jaws are not tightened sufficiently, the specimen may slip during bending.
9. Rotate the handle of the bending lever until the three rollers just touch the specimen.
Avoid over tightening - there should be no pressure between the rollers and the specimen at this stage.

5

en

R

6. Tighten the vice jaws until they hold the specimen lightly.

enR

10. Hold the bending lever firmly. Witha smooth action, swing the leveranticlockwise until it reaches its
stop position.
The specimen will now be bent
into a ‘U’ shape.
11. Swing the bending lever clockwise
until it reaches its stop position.

109

12. Release the specimen from thevice jaws and remove.
13. Inspect the coating for damage.
14. If the coating is not damaged,
select the next smallest mandrel inthe set1 and repeat steps 2 to 13
until the smallest mandrel which produces no damage to the coating is determined.

1.

6

Please note that this bending test can be also a ‘Pass/Fail’ test for a specific mandrel diameter.

93.1 PhysicalMaximum width of specimen:

76.2mm (3")

Maximum length of specimen:

80 to 100mm (3.15 to 3.93”) depending on the size of themandrel usedDimensions:

320 x 147 x 130mm(12.6 x 5.8 x 5.1")

Weight, excluding mandrels:

4 kg (9 lb)

3.2 Packaging
The Elcometer 1506-B Cylindrical Mandrel Bend Tester is packed in a cardboard and foam package.
Please ensure that this packaging is disposed of in an environmentally sensitive manner. Consult
your local Environmental Authority for further guidance.

4 MAINTENANCE
The Elcometer 1506-B Cylindrical Mandrel Bend Tester is designed to give many years reliable service
under normal operating and storage conditions.
Every 6 months, or sooner if used regularly, grease the threads and guides of the specimen vice and thebending lever roller carriage.

7en

R

3 TECHNICAL SPECIFICATIONen

R

The tester does not contain any user-serviceable components. In the unlikely event of a fault, the Elcometer
1506-B should be returned to your local Elcometer supplier or directly to Elcometer. The warranty will
be invalidated if the instrument has been opened.
Details of Elcometer offices around the world are given on the outside cover of these operating instructions.
Alternatively visit the Elcometer website, www.elcometer.com

5 ACCESSORIES
A range of mandrel sizes is available to purchase either as a complete set or as individual mandrels.
Description

Part Number

Elcometer 1506-B Set of 14 Metric Mandrels; Sizes 2 to 32mm

KT001506BP201

Elcometer 1506-B Metric Mandrel: 2mm

KT001506BF002

Elcometer 1506-B Metric Mandrel: 3mm

KT001506BF003

Elcometer 1506-B Metric Mandrel: 4mm

KT001506BF004

Elcometer 1506-B Metric Mandrel: 5mm

KT001506BF005

Elcometer 1506-B Metric Mandrel: 6mm

KT001506BF006

Elcometer 1506-B Metric Mandrel: 8mm

KT001506BF007

Elcometer 1506-B Metric Mandrel: 10mm

KT001506BF014

Elcometer 1506-B Metric Mandrel: 12mm

KT001506BF015

Elcometer 1506-B Metric Mandrel: 13mm

KT001506BF016

8Part Number

Elcometer 1506-B Metric Mandrel: 16mm

KT001506BF017

Elcometer 1506-B Metric Mandrel: 19mm

KT001506BF018

Elcometer 1506-B Metric Mandrel: 20mm

KT001506BF019

Elcometer 1506-B Metric Mandrel: 25mm

KT001506BF020

Elcometer 1506-B Metric Mandrel: 32mm

KT001506BF021

Description

Part Number

Elcometer 1506-B Set of 7 Imperial Mandrels; Sizes 1/8 to 1"

KTUS1506BP201

Elcometer 1506-B Imperial Mandrel: 1/8"

KTUS1506BF022

Elcometer 1506-B Imperial Mandrel: 1/4"

KTUS1506BF023

Elcometer 1506-B Imperial Mandrel: 3/8"

KTUS1506BF024

Elcometer 1506-B Imperial Mandrel: 1/2"

KTUS1506BF025

Elcometer 1506-B Imperial Mandrel: 5/8"

KTUS1506BF026

Elcometer 1506-B Imperial Mandrel: 3/4"

KTUS1506BF027

Elcometer 1506-B Imperial Mandrel: 1"

KTUS1506BF028

9en

R

Description

enR

6 RELATED EQUIPMENT
In addition to the Elcometer 1506-B Cylindrical Mandrel Bend Tester, Elcometer produces a wide rangeof other equipment for determining the physical characteristics of surface coatings.
Users of the Elcometer 1506-B may also benefit from the following Elcometer products:•Elcometer 1510 Conical Mandrel Bend Tester•

Elcometer 1615 Variable Impact Tester•

Elcometer 1620 Cupping Tester•

Elcometer 1542 Cross Cut Adhesion Tester

For further information contact Elcometer, your local supplier or visit www.elcometer.com

10

de1506_TMA-0322_02_04_A6L_19397_02.fm Page -1 Tuesday, August 7, 2012 9:43 AM

Elcometer 1506-B

Zylindrischer Dornbiegeprüfer

Bedienungsanleitungde1506_TMA-0322_02_04_A6L_19397_02.fm Page 0 Tuesday, August 7, 2012 9:43 AM

Rist ein eingetragenes Warenzeichen der Elcometer Limited.
Alle weiteren Markenzeichen werden anerkannt.
© Copyright Elcometer Limited. 2023.
Änderungen in der technischen Spezifikation, Ausstattung, Design und Lieferumfang ohne
Vorankündigung vorbehalten.
Alle Rechte vorbehalten. Dieses Dokument oder Teile daraus dürfen weder ganz noch teilweise ohne
vorherige schriftliche Genehmigung durch Elcometer Limited reproduziert, übertragen, verändert,gespeichert oder in eine andere Sprache übersetzt werden (weder elektronisch, mechanisch,magnetisch, optisch, manuell oder in anderer Form).
Eine Kopie dieser
www.elcometer.com.

Bedienungsanleitung

steht

auch

alsDownload

aufunserer

Webseite

Dok.Nr. TMA-0763-01 Ausgabe 01 (Based on TMA-0763 Issue 01)
Text mit Cover Nr: 31978

RINHALT
Kapitel12
3
4
56SeiteÜBER DIESE PRÜFGERÄT. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 2
PRÜFUNG EINES MUSTERS . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4TECHNISCHE DATEN . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 7PFLEGE UND WARTUNG . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 7ZUBEHÖR . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8
ÄHNLICHE PRODUKTE . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 101de1506_TMA-0322_02_04_A6L_19397_02.fm Page 1 Tuesday, August 7, 2012 9:43 AM

de1506_TMA-0322_02_04_A6L_19397_02.fm Page 2 Tuesday, August 7, 2012 9:43 AMRHerzlichen Dank das Sie sich für das Elcometer 1506-B Zylindrisches Dornbiegeprüfgerät
entschieden haben. Willkommen bei Elcometer Limited.

Elcometer ist ein weltweit führender Hersteller und Vertreiber von Oberflächen- und Betoprüfgeräten.
Elcometer Produkte sind führend in Technik, Design und Handhabung. Unsere Produkte decken alle
Bereiche der Beschichtungsprüfung ab: von der Entwicklung über die Verarbeitung bis hin zur Endkontrolle.
Der Elcometer 1506-B Zylindrischer Dornbiegeprüfer ist ein führendes Produkt auf diesem Gebiet. Mit
dem Erwerb dieses Produktes haben Sie nun Zugang zum weltweiten Servicenetzwerk von Elcometer.
Für nähere Informationen besuchen Sie bitte unsere Webseite unter www.elcometer.com

1 ÜBER DIESE PRÜFGERÄT
Der Elcometer 1506-B Dornbiegeprüfer ist ein stabiles und einfach zu handhabendes Gerät zurBestimmung der Elastizität, des Abriebs und der Dehnbarkeit von trockener Farbe auf Feinblech.
Das Feinblech wird fest in einem regulierbaren Schraubstock eingespannt und durch einen Satzverstellbarer Walzen um den Biegedorn gebogen. Zur präzisen Ermittlung, wann die Beschichtung versagt,
sind Biegedorne in verschiedenen Größen erhältlich. Siehe “ZUBEHÖR” auf seite 8.

2

R

de1506_TMA-0322_02_04_A6L_19397_02.fm Page 3 Tuesday, August 7, 2012 9:43 AM

1.1 Standards
Das Elcometer 1506-B kann in Übereinstimmung nachstehender nationaler und internationaler
Standards angewendet werden•ASTM D 522-B ersetzt ASTM D 1737•

AS/NZS 1580.402.1•

ISO 1519-2•

JIS K 5600-5-1

1.2 Lieferumfang•Elcometer 1506-B Zylindrischer Dornbiegeprüfer•

Bedienungsanleitung

Anmerkung: Biegedorne befinden sich nicht im Lieferumfang des Elcometer 1506-B undmüssen separat bestellt werden - siehe “ZUBEHÖR” auf seite 8.
Um sämtliche Vorteile Ihres neuen Elcometer 1506-B Zylindrischer Dornbiegeprüfer optimal nutzen
zu können, bitten wir Sie die nachstehende Bedienungsanleitung eingehend zu studieren. SolltenSie weitere Fragen haben so zögern Sie bitte nicht mit Elcometer, oder Ihren örtlichenHändler, in Kontakt zu treten.

31506_TMA-0322_02_04_A6L_19397_02.fm Page 4 Tuesday, August 7, 2012 9:43 AMdeR

2 PRÜFUNG EINES MUSTERS
1. Stellen Sie das Prüfgerät auf einen stabilen Tisch mit dem schwarzen Hebel nach links ausgerichtet.
2. Schwingen Sie den Biegehebel gegen den Uhrzeigersinn bis zur Stopposition.
3. Wählen Sie einen Dorn aus und setzen diesen in die dafür vorgesehene Halterung ein.
231

4. Schwingen Sie den Hebel nun in Uhrzeigerrichtung bis die Stopposition erreicht ist.

4

R

5. Öffnen Sie den integrierten Schraubstock und setzen das zu prüfende Musterblech ein. Die
beschichtete Seite muss dabei von dem Dorn abgewendet sein.
6. Schrauben Sie den Schraubstock fest, bis das Musterblech sicher fixiert ist.

7. Schieben Sie nun den Schraubstock in Richtung des Dorns, bis das Musterblech diesen berührt.
8. Ziehen Sie den Schraubstock an bis das Musterblech fest sitzt.
Ist der Schraubstock nicht richtig festgezogen, könnte das Musterblech eventuell aus der Verankerung
gerissen werden.
9. Drehen Sie nun den Hebel bis die drei Rollen das Musterblech leicht berühren.
Vermeiden Sie ein Überdrehen des Hebels! Es sollte an diesem Punkt kein Druck zwischen denRollen und dem Musterblech entstehen.
5de1506_TMA-0322_02_04_A6L_19397_02.fm Page 5 Tuesday, August 7, 2012 9:43 AM

1506_TMA-0322_02_04_A6L_19397_02.fm Page 6 Tuesday, August 7, 2012 9:43 AM

deR

10. Halten Sie den Griff fest und schwingen ihn mit
einer gleichmäβigen Bewegung entgegen dem
Uhrzeigersinn bis die Stopposition erreicht ist.

10Das Musterblech ist nun in eine U-Form
gebogen.
11. Schwingen Sie den Hebel im Uhrzeigersinn bis
die Stopposition erreicht ist.

99

12. Entfernen Sie das Musterblech aus dem
Schraubstock.
13. Prüfen
Sie
dieBeschädigungen.

Beschichtungauf

14. Sollte die Beschichtung nicht beschädigt sein,wählen Sie den nächst kleineren Prüfdorna undwiederholen die Schritte 2 bis 13 bis der kleinste Dorn der keine Beschädigung verursacht ermittelt
wurde.

a.

6Bitte beachten Sie dass diese Biegeprüfung auch als „gut/schlecht“ Test für einen bestimmten Dorndurchmesser verwendet werden kann.

Rde1506_TMA-0322_02_04_A6L_19397_02.fm Page 7 Tuesday, August 7, 2012 9:43 AM

3 TECHNISCHE DATEN
3.1 PhysikalischMax. Musterbreite:

76.2 mmMax. Musterlänge:80 mm mit 2 mm Dorn
100 mm mit 32 mm Dorn

Abmessungen:

320 mm x 147 mm x 130 mm

Gewicht (ohne Prüfdorne):

4 kg

3.2 Verpackung
DerElcometer
1506-BZylindrisches
Dornbiegeprüfer
istineinen
Pappkarton
mit
Schaumstoffeinlage verpackt. Bitte stellen Sie bei deren Entsorgung sicher dass die örtlichenEntsorgungsrichtlinien eingehalten werden. Für nähere Informationen wenden Sie sich bitte an diezuständige Behörde.

4 PFLEGE UND WARTUNG
Der Elcometer 1506-B Dornbiegeprüfer wurde so konstruiert, dass er bei normaler Beanspruchungund Aufbewahrung über viele Jahre hinweg zuverlässig arbeiten wird.
Ca. alle 6 Monate sollten die beweglichen Teile des Elcometer 1506-B (Führungen, Rollen,Schraubstock) nachgefettet werden.

7

de1506_TMA-0322_02_04_A6L_19397_02.fm Page 8 Tuesday, August 7, 2012 9:43 AM

RDas Prüfgerät besitzt keinerlei Komponenten die speziell gewartet werden müssten. In dem
unwahrscheinlichen Fall eines Defektes wenden Sie sich bitte an Elcometer oder Ihren nächsten Händler.
Die Garantie erlischt bei Öffnen des Gerätes!
Informationen über die Elcometer Niederlassungen weltweit erhalten Sie auf der Rückseite dieser
Bedienungsanleitung, oder besuchen Sie uns im Internet unter www.elcometer.com

5 ZUBEHÖR
Eine grosse Auswahl an Dornen ist sowohl einzeln, als auch im Set erhältlich.
Beschreibung

Artikelnummer

Vollständiger Satz Ersatzdorne, metrisch, 14 Stück, Größen 2mm bis 32mm

KT001506BP201

Metrisch Ersatzdorne: 2mm

KT001506BF002

Metrisch Ersatzdorne: 3mm

KT001506BF003

Metrisch Ersatzdorne: 4mm

KT001506BF004

Metrisch Ersatzdorne: 5mm

KT001506BF005

Metrisch Ersatzdorne: 6mm

KT001506BF006

Metrisch Ersatzdorne: 8mm

KT001506BF007

Metrisch Ersatzdorne: 10mm

KT001506BF014

Metrisch Ersatzdorne: 12mm

KT001506BF015

Metrisch Ersatzdorne: 13mm
8KT001506BF016

RBeschreibung

Artikelnummer

Metrisch Ersatzdorne: 16mm

KT001506BF017

Metrisch Ersatzdorne: 19mm

KT001506BF018

Metrisch Ersatzdorne: 20mm

KT001506BF019

Metrisch Ersatzdorne: 25mm

KT001506BF020

Metrisch Ersatzdorne: 32mm

KT001506BF021

Beschreibung

Artikelnummer

Vollständiger Satz Ersatzdorne, Imperial, 7 Stück, Größen 1/8 bis 1"

KTUS1506BP201

Imperial Ersatzdorne: 1/8"

KTUS1506BF022

Imperial Ersatzdorne: 1/4"

KTUS1506BF023

Imperial Ersatzdorne: 3/8"

KTUS1506BF024

Imperial Ersatzdorne: 1/2"

KTUS1506BF025

Imperial Ersatzdorne: 5/8"

KTUS1506BF026

Imperial Ersatzdorne: 3/4"

KTUS1506BF027

Imperial Ersatzdorne: 1"

KTUS1506BF028
9de

1506_TMA-0322_02_04_A6L_19397_02.fm Page 9 Tuesday, August 7, 2012 9:43 AM

1506_TMA-0322_02_04_A6L_19397_02.fm Page 10 Tuesday, August 7, 2012 9:43 AM

deR6 ÄHNLICHE PRODUKTE
Als Ergänzung des Elcometer 1506-B Zylindrischer Dornbiegeprüfer bietet Elcometer Limited eineganze Reihe weiterer Prüfgeräte zur Ermittlung der physikalischen Eigenschaften von Beschichtungen.
Anwender des Elcometer 1506-B können von nachstehenden Produkten ebenfalls profitieren:• Elcometer 1510 Konischer Dornbiegeprüfer• Elcometer 1615 Variables Kugelschlagprüfgerät• Elcometer 1620 Tiefungsprüfmaschine• Elcometer 1542 Gitterschnittprüfgerät
Für nähere Informationen wenden Sie sich bitte an Elcometer, Ihren örtlichen Händler oder besuchen Sie
uns im Internet unter www.elcometer.de10