1GS/s Digital Storage Oscilloscopes

DCS-1054B
DCS-1074B
DCS-1104B
DCS-1052B
DCS-1072B
DCS-1102B

DCS-1000B Series(1GS/s 4ch 50MHz )7” TFT WVGA color display(1GS/s 4ch 70MHz )(1GS/s 4ch 100MHz )(1GS/s 2ch 50MHz )(1GS/s 2ch 70MHz )(1GS/s 2ch 100MHz )● 100MHz/70MHz/50MHz Digital Storage Oscilloscope● Waveform update rate up to 50,000wfms/s

DCS-1052B

DCS-1054B

DCS-1072B

DCS-1074B

DCS-1102B

DCS-1104B

VerticalChannels

DCS-1xx4B : 4ch , DCS-1XX2B : 2ch+EXT

DC 〜 70MHz

Resolution

DC 〜 50MHz8 bit @1MΩ: 1mV~10V

Input Impedance

1MΩ // 16pF

DC Gain Accuracy

1mV: ±4% full scale, >2mV: ±3% full scaleBandwidth

DC 〜 100MHz

FFT: Spectral magnitude. Set FFT Vertical Scale to Linear RMS or dBV RMS, and FFTWindow to Rectangular, Hamming, Hanning, or Blackman-Harris
Trigger● 256 gradstion 7inches WVGA display

SourceCH1, CH2, CH3*, CH4*, Line, EXT***four channel models only.
**two channel models only.

Trigger Type

Edge, Pulse Width(Glitch), Video, Pulse Runt, Rise & Fall(Slope), Timeout, Alternate,Event-Delay(1~65535 events), Time-Delay(Duration, 4nS~10S), BusCoupling

AC, DC, LF rej., Hf rej., Noise rej.

Sensitivity

1div

External Trigger (two channel models only)
Range

±2.5V

Sensitivity

DC ~ 100MHz Approx. 100mV

Horizontal● 10M memory depth per channel independently● 1M FFT mathematical sampling analysis mode

Time base Range

5ns/div ~ 100s/div (1-2-5 increments) ROLL: 100ms/div ~ 100s/div

Pre-trigger

10 div maximum

Post-trigger

2,000,000 div maximum.

Real Time Sample Rate

1GSa/s max.

Record LengthMax. 10MptsAcquisition Mode

Normal, Average, Peak Detect, Single

Peak Detection

2nS (typical)

Averageselectable from 2 to 256

X-Y Mode
X-Axis Input

Channel 1; Channel 3**four channel models only

Y-Axis Input

Channel 2; Channel 4**four channel models only

Phase Shift±3° at 100kHz

Cursors and Measurement
Cursors

Amplitude, Time, Gating available; Unit: Seconds(s), Hz(1/s), Phase(degree), Ration(%)

Automatic Measurement

36 sets: Pk-Pk, Max, Min, Amplitude, High, Low, Mean, Cycle Mean, RMS, Cycle RMS,Area, Cycle Area, ROVShoot, FOVShoot, RPREShoot, FPREShoot, Frequency, Period,RiseTime, FallTime, +Width, -Width, Duty Cycle, +Pulses, -Pulses, +Edges, -Edges, FRR,FRF, FFR, FFF, LRR, LRF, LFR, LFF, Phase

Control Panel Function● Zoom In/Play and Pause Function●● X-Y Mode Display● Go/NoGo Function● Data Log Function● Digital Filter Function● 36 Measurement Parameter Selections● Serial Bus Decode and Trigger

AutosetSingle-button, automatic setup of all channels for vertical, horizontal and trigger
systems, with undo Autoset

Save Setup

20set

Save Waveform

24setDisplay
TFT LCD Type

7”TFT WVGA color display 800 horizontal × 480 vertical pixels (WVGA)

Waveform Update Rate

50,000 waveforms per second, maximumInterfaceUSB PortUSB 2.0 High-speed host port X1, USB High-speed 2.0 device port X1

Ethernet Port (LAN)RJ-45 connector, 10/100Mbps with HP Auto-MDIX (4ch Model Only)

GeneralDimensions

384mm X 208mm X 127.3mm

Weight2.8kg

Power Source

100V~240V AC, 50Hz~60Hz, Auto selection, Power consumption: 30 Watts

7F Towa Fudosan Shin Yokohama Bldg.,2-18-13 Shin Yokohama, Kohoku-ku, Yokohama, Kanagawa, 222-0033 Japan
TEL: +81(0)45-620-2305 FAX: +81(0)45-534-7181

https://www.texio.co.jp/