Spectrum Analyzer
GSP-930PROGRAMMING MANUALGW INSTEK PART NO. 82SP-93000M01

ISO-9001 CERTIFIED MANUFACTURER

This manual contains proprietary information, which is protected bycopyright. All rights are reserved. No part of this manual may be
photocopied, reproduced or translated to another language without
prior written consent of Good Will company.
The information in this manual was correct at the time of printing.
However, Good Will continues to improve products and reserves therights to change specification, equipment, and maintenanceprocedures at any time without notice.
Good Will Instrument Co., Ltd.
No. 7-1, Jhongsing Rd., Tucheng Dist., New Taipei City 236, Taiwan.

エラー! スタイルが定義されていません。

Table of Contents
SAFETY INSTRUCTIONS ............................................................. 2
GETTING STARTED ...................................................................... 7
GSP-930 Introduction..................................8
Accessories ................................................. 10
Appearance .................................................. 12
REMOTE CONTROL ....................................................................23
Interface Configuration............................. 24
Command Syntax........................................ 34
Status Registers......................................... 38
Command List ............................................. 49

1

エラー! スタイルが定義されていません。

SAFETY INSTRUCTIONSThis chapter contains important safety instructions
that you must follow during operation and storage.
Read the following before any operation to insure
your safety and to keep the instrument in the best
possible condition.

Safety Symbols
These safety symbols may appear in this manual or on the instrument.

WARNING

Warning: Identifies conditions or practices that could
result in injury or loss of life.

CAUTION

Caution: Identifies conditions or practices that could
result in damage to the instrument or to otherproperties.
DANGER High Voltage
Attention Refer to the Manual
Earth (ground) Terminal
Frame or Chassis Terminal2エラー! スタイルが定義されていません。
Do not dispose electronic equipment as unsorted
municipal waste. Please use a separate collection
facility or contact the supplier from which this
instrument was purchased.

Safety Guidelines
General Guideline • Do not place any heavy object on the instrument.
CAUTION•

Avoid severe impact or rough handling that leads todamaging the instrument.

•

Do not discharge static electricity to the
instrument.

•

Use only mating connectors, not bare wires, for theterminals.

•

Ensure signals to the RF input do not exceed+30dBm.

•Ensure reverse power to the TG output terminal
does not exceed +30dBm.

•

Do not supply any input signals to the TG output.

•

Do not block the cooling fan opening.

•

Do not disassemble the instrument unless you arequalified.

(Measurement categories) EN 61010-1:2010 specifies themeasurement categories and their requirements as follows. The
instrument falls under category II.
• Measurement category IV is for measurement performed at thesource of low-voltage installation.
• Measurement category III is for measurement performed in thebuilding installation.
• Measurement category II is for measurement performed on thecircuits directly connected to the low voltage installation.
• Measurement category I is for measurements performed on
circuits not directly connected to Mains.

3

GSP-930 User Manual
Power Supply
WARNING

Battery
CAUTION
Cleaning

Operation
Environment•

AC Input voltage range: 100V~240V•

Frequency: 50/60Hz•

To avoid electrical shock connect the protective
grounding conductor of the AC power cord to an
earth ground.

•

Rating: 10.8V, 6 cell Li-ion battery•

Turn off the power and remove the power cord
before installing or removing the battery.

•

Disconnect the power cord before cleaning.

•

Use a soft cloth dampened in a solution of mild
detergent and water. Do not spray any liquid.

•

Do not use chemicals containing harsh material
such as benzene, toluene, xylene, and acetone.

•

Location: Indoor, no direct sunlight, dust free,almost non-conductive pollution (Note below)•

Temperature: 5°C to 45°C•

Humidity: <90%

(Pollution Degree) EN 61010-1:2010 specifies the pollution degrees
and their requirements as follows. The instrument falls under
degree 2.
Pollution refers to “addition of foreign matter, solid, liquid, or
gaseous (ionized gases), that may produce a reduction of dielectricstrength or surface resistivity”.
• Pollution degree 1: No pollution or only dry, non-conductive
pollution occurs. The pollution has no influence.
• Pollution degree 2: Normally only non-conductive pollution
occurs. Occasionally, however, a temporary conductivity causedby condensation must be expected.
• Pollution degree 3: Conductive pollution occurs, or dry, nonconductive pollution occurs which becomes conductive due tocondensation which is expected. In such conditions, equipment
is normally protected against exposure to direct sunlight,
precipitation, and full wind pressure, but neither temperature nor
humidity is controlled.

4エラー! スタイルが定義されていません。
Storageenvironment

Disposal•

Location: Indoor•

Temperature: -20°C to 70°C•

Humidity: <90%

Do not dispose this instrument as unsorted municipal
waste. Please use a separate collection facility or
contact the supplier from which this instrument waspurchased. Please make sure discarded electricalwaste is properly recycled to reduce environmental
impact.

5

GSP-930 User Manual

Power cord for the United Kingdom
When using the instrument in the United Kingdom, make sure the power
cord meets the following safety instructions.
NOTE: This lead/appliance must only be wired by competent persons
WARNING: THIS APPLIANCE MUST BE EARTHEDIMPORTANT: The wires in this lead are coloured in accordance with thefollowing code:
Green/ Yellow:
Earth
Blue:
NeutralBrown:
Live (Phase)

As the colours of the wires in main leads may not correspond with thecoloured marking identified in your plug/appliance, proceed as follows:
The wire which is coloured Green & Yellow must be connected to the
Earth terminal marked with either the letter E, the earth symbolorcoloured Green/Green & Yellow.
The wire which is coloured Blue must be connected to the terminal
which is marked with the letter N or coloured Blue or Black.
The wire which is coloured Brown must be connected to the terminal
marked with the letter L or P or coloured Brown or Red.
If in doubt, consult the instructions provided with the equipment or
contact the supplier.
This cable/appliance should be protected by a suitably rated andapproved HBC mains fuse: refer to the rating information on theequipment and/or user instructions for details. As a guide, a cable of
0.75mm2 should be protected by a 3A or 5A fuse. Larger conductors
would normally require 13A types, depending on the connection method
used.
Any exposed wiring from a cable, plug or connection that is engaged ina live socket is extremely hazardous. If a cable or plug is deemedhazardous, turn off the mains power and remove the cable, any fuses
and fuse assemblies. All hazardous wiring must be immediately
destroyed and replaced in accordance to the above standard.

6エラー! スタイルが定義されていません。

GETTING STARTEDThis chapter provides a brief overview of the GSP930, the package contents, instructions for first time
use and an introduction to the front panel, rear
panel and GUI.

GSP-930 Introduction .................................................................. 8
Main Features ...............................................................................8

Accessories ..................................................................................10
Package Contents.................................................................... 11

7

GSP-930 User Manual

GSP-930 Introduction
The GSP-930 is the most advanced spectrum analyzer GW Instek has
produced to date. The GSP-930 features a split window display to view
data in spectrum, topographic or spectrographic views.

Main Features
Performance

Features8•

9kHz~3GHz bandwidth•

1Hz resolution•

Nominal RBW accuracy of 5% <750kHz, 8%@>750kHz•

Video bandwidth 1Hz~1MHz (10 steps)•

Amplitude measurement range: DANL~30dBm(frequency dependent)•

Input attenuation: 0 ~ 50dB•

Phase noise: < -88dBc/Hz@1GHz, 10kHz•

10%-step increments for RBW bandwidth•Three display modes: Spectrum, Topographic andSpectrographic•

Split window display•

Built-in EMI filter•

Auto Wake-up•

Built-in preamplifier•

Gate sweep•

Marker Frequency counter•

Two operating modes: Spectrum and Power Meter
mode•

SEM measurement•

ACPR measurementエラー! スタイルが定義されていません。

Interface•

OCBW measurement•

Channel power measurement•

Demodulation analyzer•

Diverse marker functions and features with Peak
Table

•

Sequence function to automatically perform preprogrammed sequential operations•

Optional battery operation•

8.4 color LCD (800×600)•

On-screen menu icons•

DVI-I video output•

RS-232 with RTS/CTS hardware flow control•

USB 2.0 with support for USB TMC•

LAN TCP/IP with LXI support•

Optional GPIB/IEEE488 interface•

IF output @ 886MHz•

Headphone output•

REF (reference clock) input/output BNC ports•

Alarm/Open collector output BNC port•

Trigger/Gate input BNC ports•

RF N-type input port•

Tracking generator output•

DC +7V/500mA output SMB port

9

GSP-930 User Manual

Accessories
Standard
Accessories

Options

Optional
Accessories

10

Part number

Description

Region dependantUser manual

Region dependantPower cord

Option number

Description

Opt1.

Tracking generator

Opt2.

Battery (11.1V/5200mAH
Li-ion battery)

Opt3.

GPIB interface (IEEE 488
bus)Part number

Description

PWS-06

USB Average Power
Sensor (up to 6200 MHz;
-32 to 20 dBm)

GRA-415

6U Rack mount kitエラー! スタイルが定義されていません。

Package Contents
Check the contents before using the GSP-930.
Opening the box

Contents(single unit)•

Main unit(may include optional
GPIB, TG output)•

Quick Start manual•

User Manual CD•

Power cord x1 (region
dependent)•

Optional battery pack•

Calibration certificate

11GSP-930 User Manual

Appearance
GSP-930 Front PanelLCD display

Function keys

Main keys

12

800×600 color LCD display. The display shows thesoft keys for the current function, frequency,
amplitude and marker information.
F1

~

F7

The F1 to F7 function keys directly
correspond to the soft keys on theright-hand side of display.
Sets the center frequency, startfrequency, stop frequency, center
frequency step and frequency offset
values.

エラー! スタイルが定義されていません。
Sets the span, with options for fullspan, zero span and last span.
Sets the amplitude reference level,
attenuation, pre-amplifier controls,
scale and other options forattenuation and scale.
Automatically searches the peaksignal with maximum amplitude anddisplays it with appropriate
horizontal and vertical scales.
Control keys

Sets the resolution bandwidth, video
bandwidth, average type and turns
the EMI filter on/off.
Sets the sweep time and gate time.

Sets traces and trace related
functions.
The Display key configures thewindowing mode and basic display
properties.
Accesses measurement options
such as ACPR, OCBW, demodulation
measurements, SEM, TOI and otheradvanced measurements.
Sets and tests Pass/Fail limit lines.

Access, set and edit program
sequences.
13

GSP-930 User ManualSets the triggering modes.

File

File utilities options

The Quick Save utility allows you tosave either the state, trace, screen
limit line, correction or sequencewith only a single press.
Save the trace, state etc., and save
options.
Recall the trace, state etc., andrecall options.
MarkerTurns the Markers on/off andconfigures the markers.
The Marker-> key positions themarkers on the trace.
Finds each maximum and minimum
peak. Used with the Marker function.

State

The Preset key will restore thespectrum analyzer to the Factory or
User-defined settings.
The Preset key will also return the
instrument back to local control
after it has been in remote control
mode.

14

エラー! スタイルが定義されていません。
The Mode key sets the spectrum
analyzer to either Spectrum orPower Meter mode.
The System key shows system
information, settings and othersystem related functions.
Power key

Turns the instrument on/off.

Scroll wheel

Edit values, select listed items.

Arrow keys

Increment/decrement values (insteps), select listed items.

RF input terminal

RF input port. Accepts RF inputs.
RF INPUT 50
DC ±50VMAX.
+30dBm MAX.

DC power supply
500mA MAX.
DC 7V
OUTPUT•

Maximum input: +33dBm•

Input impedance: 50Ω•

Maximum DC voltage: ±50V•

N-type: female

SMB port supplies power for optional
accessories.
•

DC +7V•

500mA Max.

15

GSP-930 User Manual
Numeric keypad
7

8

9

4

5

6

1

2

3

BK SP

/

Enter

0

TG output port

USB A, Micro SD

16

The numeric keypad
is used to enter
vales andparameters. It is
often used inconjunction with thearrow keys and scroll
wheel.

The Tracking Generator (TG) output
source.
•N-type: female•

Input impedance: 50Ω•

Output power: -50dBm to 0dBm•

Maximum reversed power:+30dBmUSB A port, Micro SD port forsaving/recalling settings/files.

エラー! スタイルが定義されていません。

Rear Panel

RS232

RS232 9 pin DSUB port.

IF OUT

SMA IF Out port.

DVI-I

DVI video out port. Supports SVGA
(800X600) @ 60Hz.

FanPower Socket

Power Socket:
100~240V, 50/60Hz.

17

GSP-930 User Manual
Battery packVoltage: 10.8V
Capacity: 5200mAH

REF INBNC female reference input.

REF OUT

BNC female reference output:
10MHz, 50Ω impedance

Security Lock
ALARM OUT

BNC female open collector Alarm
output.

TRIG IN/GATE IN

BNC female 3.3V CMOS trigger
input/gated sweep input.

Phone

3.5mm stereo headphone jack (wired
for mono operation)

USB B

USB B Device port. USB 1.1/2.0

LAN

RJ-45 10Base-T/100Base-Tx

18

エラー! スタイルが定義されていません。

Display

Reference level

Displays the reference level.

Attenuation

Displays the vertical scale (attenuation) of the input
signal.

Date/Time

Displays the date and time.

Markerinformation

Displays marker information.

LXI iconThis icon indicates the status of the LXI connection.
For details, see page 25.

Function menu

Soft menu keys associated with the F1 to F7 function
keys to the right of the display.

19

GSP-930 User Manual
Sweep settings

Sweep icon that shows the sweep
status.

Trigger settings

Trigger icon that shows the trigger
status.

Pre-amp settings

Pre-amplifier icon that shows the
Pre-amplifier status.

Trace anddetection
settings

Trace icon that shows the trace type
and the detection mode used for eachtrace.

Status Icons

Displays the interface status, power source status,
and alarm status, etc. See the Status Icon Overview
on page 21 for a list of the status icons.

Frequency/
Bandwidth
settings

Displays the Start, Center and Stop frequencies,
RBW, VBW, Span and Sweep settings.

Entry/Message
area

This area is used to show system messages, errors
and input values/parameters.

Trace andwaveforms

Main display showing the input signals, traces, limit
lines and marker positions.

20

エラー! スタイルが定義されていません。

Status Icon Overview
PreAmp

Indicates that the pre amplifier is on.

ACShown when running on AC power.

AC Charge

Shown when the AC power is charging
the battery.

Alarm Off

Alarm buzzer output is currently off.

Alarm On

Alarm buzzer output is currently on.

Amplitude Offset

Indicates that the amplitude-shift is
active. This icon appears when
amplitude-related functions are used:
Reference level offset
Amplitude Correction
Input Z = 75Ω
Input Z cal >0

Battery indicator~

Indicates the battery charge.

Bandwidth
Indicator

Indicates that the RBW or VBW
settings are in manual mode.

Average

Indicates that the Average function is
active.

External Lock

Indicates that the system is now
locked and refers to the externalreference input signal

21

GSP-930 User Manual
External Trigger

External trigger signal is being used.

Math

Trace math is being used.

Sequence
Indicator

Shown when a sequence is running.

Sweep Indicator

Indicates that the sweep time ismanually set.

Tracking
generator

Indicates the tracking generator isturned on.

TG Normalization

Indicates that the tracking generator
has been normalized.

Wake-up clock

Indicates that the wake-up clock isturned on.

USB

Indicates that a USB flash drive isinserted into the front panel and isrecognized.

Micro SD

Indicates that a micro SD card isinserted into the front panel and isrecognized.

22エラー! スタイルが定義されていません。

REMOTE CONTROLThis chapter describes the basic configuration ofIEEE488.2 based remote control. This chapter
includesinterface configuration, a remote control
overview as well as the control syntax andcommands.

Interface Configuration ..............................................................24
Command Syntax .........................................................................34
Status Registers ..........................................................................38
Command List...............................................................................49

23

GSP-930 User Manual

Interface Configuration
Configure to USB Remote InterfaceUSB
configuration

Panel operation

PC sideconnector

Type A, host

GSP sideconnector

Rear panel Type B, slave

Speed

1.1/2.0 (full speed/high speed)

USB Class

USB TMC (USB T&M class)

1. Connect the USB cable to the rear
panel USB B port.
2. Press>More 1/2[F7]>RmtInterface
Config[F2]>USB Mode and toggle the USB mode
to Device.

Note

It may take a few moments to switch USB modes.

Configure GPIB Interface
To use GPIB, the optional GPIB port must be installed.
Configure GPIB

3. Ensure the spectrum anlayzer is off before
proceeding.
4. Connect a GPIB cable from a
GPIB controller to the GPIB porton the spectrum analyzer.
5. Turn the spectrum analyzer on.

24

エラー! スタイルが定義されていません。

6. Press>More 1/2[F7]>RmtInterface
Config[F2]>GPIB Addr and set the GPIB address.
GPIB address

0~30

GPIB constraints • Maximum 15 devices altogether, 20m cable length,•••

2m between each device
Unique address assigned to each device
At least 2/3 of the devices turned On
No loop or parallel connection

Configure the LAN and LXI Interface
The GSP-930 is a class C LXI compliant instrument. The LXI
specification allows instrumentation to be configured for basic remotecontrol or monitoring over a LAN.
For details on the LXI specification and compliance classes, please see
the LXI website @ http://www.lxistandard.org.
Background

The LAN interface is used for remote control overa network. The spectrum analyzer supports DHCP
connections so the instrument can be
automatically connected to an existing network.
Alternatively, network settings can also be
manually configured.

LAN configuration
Settings

IP Address

Default Gateway

Subnet Mask

DNS Server

DHCP on/off
Connection

Connect an Ethernet cable from the
network to the rear panel LAN port.

25

GSP-930 User Manual
Settings

7. Press>More 1/2[F7]>RmtInterface>
LAN[F2]>LAN Config[F1] to set the LAN settings:

IP Address[F1] Sets the IP address.
Subnet Mask[F2] Sets the subnet mask.
Default
Gateway[F3]Sets the default gateway.
DNS Server[F4] Sets the DNS server address
LAN Config[F5] Toggles the LAN configuration
between DHCP and manual
settings.
8. Press Apply[F6] to confirm the LAN configuration
settings.
Display Icon

The LXI icon turns green when connected
to a LAN and will flash if the“Identification” setting is on, see page 30.

Set Password

The password on the LXI webpage can be set fromthe spectrum analyzer. The password is shown inthe system information.
By default the password is set to: lxiWNpwd>More 1/2[F7]>RmtInterface
9. Press
Config[F2]>LAN[F2]>LXIPassword[F2] to set the
password.
10. Enter the password using the F1~F7
keys, as shown below, or use thenumeric keypad to enter numbers:

Limitations:••26

No spaces
Only 1~9, A~Z, a~z characters allowedエラー! スタイルが定義されていません。
A
B

LXI
Password>

C

a

ABCDE

D

b

FGHIJ

E

c

KLMNO

Lowercase

d

PQRST

ReturneUVWXYZ

ReturnReturnMenu tree to enter the password
11. The password appears on the bottom of thescreen as it is created.

12. Press
Reset LAN

to confirm setting the password.

It may be necessary to reset the LAN
configuration settings before the LAN can be used.
13. Press>More 1/2[F7]>RmtInterface
Config[F2]>LAN Reset[F3] to reset the LAN.
14. The GSP-930 will now automatically reboot.

Note

Each time the LAN is reset, the default password
is restored.
Default password: lxiWNpwd

27

GSP-930 User Manual

Configure RS232C
Background

The RS232C interface is used for remote control
with a PC.

RS232C
Configuration
settings

Baud Rate

Stop bit: 1 (fixed)

Parity: none (fixed)

Data bit: 8 (fixed)

Connection

Connect an RS232C cable fromthe PC to the rear panel RS232
port.
15. Press>More 1/2[F7]>RmtInterface
Config>RS232 BaudRate[F4] to set the baud rate.
300
2400
19200
115200

600
4800
38400

1200
9600
57600

RS232C Remote Control Function Check
Functionality
check

Invoke a terminal application such as MTTTY(Multi-Threaded TTY).
To check the COM port No, see the Device
Manager in the PC. For WinXP; Control panel →System → Hardware tab.
Run this query command via the terminal after the
instrument has been configured for RS232 remotecontrol (page 26).
*idn?

28

エラー! スタイルが定義されていません。
This should return the Manufacturer, Model
number, Serial number, and Firmware version in
the following format.
•

GW-INSTEK,GSP-930, XXXXXXXXXXXX, V.X.X.X.X
Manufacturer: GW-INSTEKModel number : GSP-930
Serial number : XXXXXXXXXXXX
Firmware version : V.X.X.X

Note

For further details, please see the programming
manual, available on the GW Instek web site @www.gwinstek.com.

29

GSP-930 User Manual

LXI Browser Interface and Function Check
Functionality
checkEnter the IP address of the spectrum analyzer in a
web browser after the instrument has been
configured and connected to the LAN (page 25).
http:// XXX.XXX.XXX.XXX
The web browser interface appears:

Welcome Page

Note

30

The Welcome Page lists all the LXI and LAN
configuration settings as well as the instrument
identification. The instrument identification can bedisabled from this page.

The LXI icon in the GSP-930 display will
flash when the Identification setting isturned on.

エラー! スタイルが定義されていません。
View & Modify
Configuration

The View & Modify Configuration allows you tomodify the LAN settings from the browser. A
password must be entered to alter the settings.
Password: lxiWNpwd[Note: password is case sensitive.]

Note

If the “Factory Defaults” option is chosen, thepassword will be reset back to the default
password
It will also be necessary to manually reset the
spectrum analyzer when a message prompts you
to do so on the web browser.

31

GSP-930 User Manual
SCPI Command

The SCPI Command page allows you to enter
SCPI commands directly from the browser for full
remote control. Please see the programming
manual for details. A password must be entered
before remote commands can be used.
Password: lxiWNpwd[Note: password is case sensitive.]

Get Image

Note

32

The Get Image page allows the browser toremotely capture a screenshot of the GSP-930
display.

For further details, please see the programming
manual, available on the GW Instek web site @www.gwinstek.com.

エラー! スタイルが定義されていません。

GPIB/LAN Control Function Check
Functionality
check

Please use the National Instruments Measurement& Automation Controller software to confirm
GPIB/LAN functionality.
See the National Instrument website,
http://www.ni.com for details.

Note

For further details, please see the programming
manual, available on the GW Instek web site @www.gwinstek.com.

33

GSP-930 User Manual

Command Syntax
Compatible
Standard
Command
Structure

IEEE488.2

Partial compatibility

SCPI, 1999

Partial compatibility

SCPI commands follow a tree-like structure,organized into nodes. Each level of the command
tree is a node. Each keyword in a SCPI command
represents each node in the command tree. Eachkeyword (node) of a SCPI command is separated
by a colon (:).
For example, the diagram below shows an SCPI
sub-structure and a command example.

Command types

There are a number of different instrumentcommands and queries. A command sends
instructions or data to the unit and a query
receives data or status information from the unit.
Command types

34

SimpleA single command with/withouta parameter

Example*IDN?

エラー! スタイルが定義されていません。

Query

A query is a simple or compoundcommand followed by a question
mark (?). A parameter (data) isreturned.

ExampleCALCulate:CSO:STATe?

CompoundTwo or more commands on thesame command line. Compoundcommands are separated witheither a semi-colon (;) or asemi-colon and a colon (;:).
A semi-colon is used to join two
related commands, with the
caveat that the last commandmust begin at the last node of
the first command.
A semi-colon and colon areused to combine two commands
from different nodes.

ExampleCommand Forms:calc:acpr:stat?;:calc:cso:stat?

Commands and queries have two different forms,
long and short. The command syntax is writtenwith the short form of the command in capitals
and the remainder (long form) in lower case.
The commands can be written in capitals or
lower-case, just so long as the short or long forms
are complete. An incomplete command will not berecognized.
Below are examples of correctly writtencommands.
35

GSP-930 User Manual
Long
form
Short
form
Square Brackets

CALCulate:ACPR:STATe?
calculate:acpr:state?
CALCULATE:ACPR:STATE?
CALC:ACPR:STAT?
calc:acpr:stat?

Commands that contain square brackets indicatethat the contents are optional. The function of thecommand is the same with or without the square
bracketed items, as shown below.
Both “:OUTPut[:STATe]?” and “OUTPut” areboth valid forms.

Command Format

1.
2.
3.
Commonparameters

36

Command header
Space
Parameter 1

4.
5.

Comma (no space
before/after comma)
Parameter 2

Type

Description

Example<Boolean>

Boolean logic

0, 1

<NR1>

integers

0, 1, 2, 3

<NR2>

decimal numbers 0.1, 3.14, 8.5<NR3>

floating point<NRf>

any of NR1, 2, 3 1, 1.5, 4.5e-1<freq>

NR3 + unit

<limit num>

NR1<point>NR1<offset>NR3 + unit

30 db

<rel_ampl>

NR3 + unit

30 db

4.5e-1, 8.25e+1

2.5 mhz

エラー! スタイルが定義されていません。
<ampl>

NR3 +<trace name> NR1

trace1<time>

2.3e-6 ms

NR3 + unit

<ip address> string
Message
Terminator

30 mv

LF

172.16.20.20

Line feed code37

GSP-930 User Manual

Status Registers
Status Registers Overview
Description

The status registers are used to determine thestatus of the spectrum analyzer. The status
registers maintain the status of the pass/faillimits, trigger status and other operation statuses.
The status registers are arranged in a number ofgroups:
•
•
•
•
•
•
•

38

Questionable Status Registers
Standard Event Status Registers
Operation Status Registers
Status Byte Register
Service Request Enable Register
Error/Event Queue
Output Bufferエラー! スタイルが定義されていません。
Questionable
Status ACP Limit
Register

Questionable
Status SEM
Limit Register

Questionable
Status TOI Limit
Register

15

15

15

15

15

14

14

14

14

14

13

13

13

13

12

12

12

11

11

11

11

10

10

Not usedNot used

10

9

9

9

8

8

8

7

7

7

6

6

6

5

Wait fortrigger

5

4

Measuring

4

4

3

Sweeping

3

3

2

2
Not used0

5

2

Not used

9
8
7
6
5
4
3
2

1

1

Main Channel
Low Fail

100Main ChannelHigh Fail

0

12
11
10

Offset 5
Lower Fail
Offset 5Upper Fail
Offset 4
Lower Fail
Offset 4
Upper Fail
Offset 3
Lower Fail
Offset 3Upper Fail
Offset 2
Lower Fail
Offset 2Upper Fail
Offset 1
Lower Fail
Offset 1
Upper Fail98

Not used

7
6
5
4
3
2
1
0

3rd Lower
Fail
3rd Upper
Fail

Questionable
Status Register

15
14

+

+

1

Invalidspan of BW

Adj4 Low
Fail
Adj4 HighFail
Adj3 Low
Fail
Adj3 HighFail
Adj2 Low
Fail
Adj2 HighFail
Adj1 Low
Fail
Adj1 HighFail13

Not used

12

+

10

Not used

+

+

+

13
12

TOI Limit

11

SEM Limit

10

ACP Limit

9

Limit Fail

8

Uncal

7

SRE

STB

ESE

7

&

7

Status:Operation

7

&

6

&

6

RQS/MSS

6

&

5

&

5

ESB

5

&

5

4

&

4

MAV

4

&

4

3

&

3

Status:Questionable

3

&

3

Devicedependent error

1

2

&

2

Error/Event
Summary2&

2

Query Error

0

1

&

1

1

&

1

0

&

0

0

&

0

Requestcontrol
Operationcomplete

Not usedNot used
Pmet Limit
Fail

ESR

6

7

Power on

5

6

User
RequestCommanderror
Execution
Error

4

Not used
Frequency

32

Not used+

Questionable
StatusFrequency
Register

Operation Status
Register

Output Buffer

Error/Event
Queue

39

GSP-930 User Manual
Status Register
Structure

Each status register (excluding the status byte
register) is divided into a number of register
structures:
•
•
•
•
•

Condition register
Positive transition register
Negative transition register
Event Register
Event Enable Register

Condition
Registers

The condition registers report the state of the
GSP-930. Condition registers can only be read.

PTR Registers

The positive transistion registers are used to filter
for events that occur from a negative to a positivetransition.

NTR Registers

The negative transition registers are used to filter
for events that occur from a positive to negativetransistion.

Event Registers

The PTR/NTP registers dictate the typetransistion conditions that will set thecorresponding bits in the event registers. The
event registers can only be read. Reading an event
register will clear it.

Event Enable
Registers

The event enable registers determine which
events in the corresponding event registers will
set the summary bits in a higher-order register.

40

エラー! スタイルが定義されていません。

Status Byte Register (STB)
Overview

The Status Byte register consolidates the status
events of all the status registers. The Status Byte
register can be cleared with the *CLS command.
Any bits set in the Status byte register acts as asummary register for all the other status registers
and indicates if there is a service request, an error
in the Error Queue or data in the Output Queue.
Reading the Status Byte register will reset theregister to 0.
The Service Request Enable Register controls
which bits in the Status Byte Register are able togenerate service requests.

Bit Summary

Bit
2

3

4

Weight Description
4
Error/Event Queue Summary bit: Thisbit is set when there is a message inthe error queue.
8
Questionable Status Summary Bit: Thisis the summary bit for the Questionable
Status Register.
16
MAV: This bit is set when there is amessage in the output queue.
41

GSP-930 User Manual
5

32

6

64

7

128

ESB: This is the summary bit for theStandard Event Register.
MSS/RQS: The MSS bit is the summary
bit for the Service Request Enable
Register. The RQS bit is set to 1 when
the MSS bit is set to 1.
Operation Status Summary Bit: This is
the summary bit for the Operation
Status Register.

Standard Event Status Register (ESR)
Overview

Bit Summary

The Standard Event Status Register Group
indicates if any errors have occurred or fail limits
tripped. Reading this register will clear the register.
Bit
2
3

4
5
6
7

42

Weight Description
4
Query Error: When a query error has
occurred, this bit is set to 1.
8
Device-Specific Error: When a device
dependent error has occurred, this bit
is set to 1.
16
Execution Error: When an execution
error has occurred, this bit is set to 1.
32
Command Error: When a command error
has occurred, this bit is set to 1.
64
User Request: When a panel key ispressed, this bit is set to 1.
128
Power On: When the instrument isturned off Æ on, this bit is set to 1.

エラー! スタイルが定義されていません。

Operation Status Register
Overview
Bit Summary

The Operation Status Register Group indicatesthe operating status of the GSP-930.
Bit
3
4
5

Weight Description
8
Sweeping: Indicates that a sweep is inprogress.
16
Measuring: The instrument is currently
performing a measurement.
32
Waiting for Trigger: The instrument is ina “wait for trigger” state.

Questionable Status Register
Overview
Bit Summary

The Questionable Status Register Group indicates
if any limits have been tripped.
Bit
5

8

9
10

11

12

Weight Description
32
Frequency Status Summary Bit: This is
the summary bit of the Frequency
Status Register.
256
Uncal: The bit is set when an signal
level occurs because the sweep is too
fast
512
Limit fail: This bit is set to 1 when thelimit line has been violated.
1024
ACP Limit Status Summary Bit: This is
the summary bit for the ACP Limit
Status Register.
2048
SEM Limit Status Summary Bit: This is
the summary bit for the SEM Limit
Status Register.
4096
TOI Limit Status Summary Bit: This is
the summary bit for the TOI Limit
Status Register.
43

GSP-930 User Manual
13

44

8192

Pmet Limit Fail: This bit is set to 1
when the power meter limit has been
violated.

エラー! スタイルが定義されていません。

Questionable Status Frequency Register
Overview
Bit Summary

The Questionable Status Frequency Register
indicates if the span or BW settings are invalid.
Bit
5

Weight Description
32
Invalid Span or BW: This is bit is set to1 when there is an invalid span or
bandwidth (setting) during thefrequency count.

Questionable Status ACP Limit Register
Overview

Bit Summary

The Questionable Status ACP Limit Register
Group indicates if any adjacent channel limits havebeen tripped.
Bit
0

1

2
3
4
5
6
7

Weight Description
1
Main Channel High Fail: This bit is set
to 1 when the Main CH HLimit has been
violated.
2
Main Channel Low Fail: This bit is set to1 when the Main CH LLimit has been
violated.
4
Adj1 High Fail: This bit is set to 1 whenthe ADJCH 1 HLimit has been violated.
8
Adj1 Low Fail: This bit is set to 1 whenthe ADJCH 1 LLimit has been violated.
16
Adj2 High Fail: This bit is set to 1 whenthe ADJCH 2 HLimit has been violated.
32
Adj2 Low Fail: This bit is set to 1 whenthe ADJCH 2 LLimit has been violated.
64
Adj3 High Fail: This bit is set to 1 whenthe ADJCH 3 HLimit has been violated.
128
Adj3 Low Fail: This bit is set to 1 whenthe ADJCH 3 LLimit has been violated.
45

GSP-930 User Manual

Questionable Status SEM Limit Register
Overview

46

The Questionable Status SEM Limit Register
Group indicates if any of the SEM offset limits
have been tripped.

エラー! スタイルが定義されていません。
Bit Summary

Bit
0

1

2

3

4

5

6

7

8

9

Weight Description
1
Offset 1 Upper Fail: This bit is set to 1
when the XXX upper limit has been
violated.
2
Offset 1 lower Fail: This bit is set to 1
when the XXX lower limit has been
violated.
4
Offset 2 Upper Fail: This bit is set to 1
when the XXX upper limit has been
violated.
8
Offset 2 lower Fail: This bit is set to 1
when the XXX lower limit has been
violated.
16
Offset 3 Upper Fail: This bit is set to 1
when the XXX upper limit has been
violated.
32
Offset 3 lower Fail: This bit is set to 1
when the XXX lower limit has been
violated.
64
Offset 4 Upper Fail: This bit is set to 1
when the XXX upper limit has been
violated.
128
Offset 4 lower Fail: This bit is set to 1
when the XXX lower limit has been
violated.
256
Offset 5 Upper Fail: This bit is set to 1
when the XXX upper limit has been
violated.
512
Offset 5 lower Fail: This bit is set to 1
when the XXX lower limit has been
violated.

47

GSP-930 User Manual

Questionable Status TOI Limit Register
Overview

Bit Summary

The Questionable Status TOI Limit Register Group
indicates if the 3rd Order Upper or Lower limit has
been tripped.
Bit
0

1

48

Weight Description
1
3rd Upper Fail: This bit is set to 1 whenthe 3rd Order Upper limit has been
tripped.
2
3rd Lower Fail: This bit is set to 1 whenthe 3rd Order Lower limit has been
tripped.

エラー! スタイルが定義されていません。

Command List
SCPI Commands *CLS ............................................................................................. 60*IDN............................................................................................... 60*ESE.............................................................................................. 61*ESR? ........................................................................................... 61*OPC............................................................................................. 61*RST ............................................................................................. 62*SRE.............................................................................................. 62*STB? ........................................................................................... 62*TST.............................................................................................. 62*WAI .............................................................................................. 63

CALCulateCommands:CALCulate:ACPR:ACHannel<n>:HLIMit:FAIL? ............ 65:CALCulate:ACPR:ACHannel<n>:LLIMit:FAIL?............. 66:CALCulate:ACPR:ACHannel<n>:LOWer? ...................... 66:CALCulate:ACPR:ACHannel<n>:UPPer?....................... 66:CALCulate:ACPR:ACHannel<n>:STATe ........................ 67:CALCulate:ACPR:CHANnel:HLIMit:FAIL? ..................... 67:CALCulate:ACPR:CHAN:LLIMit:FAIL?............................ 68:CALCulate:ACPR:CHPower?.............................................. 68:CALCulate:ACPR:STATe..................................................... 68:CALCulate:CNR:RESult?...................................................... 69:CALCulate:CNR:STATe........................................................ 69:CALCulate:CSO:RESult?...................................................... 69:CALCulate:CSO:STATe........................................................ 70:CALCulate:CTB:RESult? ...................................................... 70:CALCulate:CTB:STATe........................................................ 70:CALCulate:CTB:RESTart..................................................... 71:CALCulate:DELTamarker<n>:PAIR:SPAN..................... 71:CALCulate:DELTamarker<n>:PAIR:CENTer................. 71:CALCulate:DELTamarker<n>:X ......................................... 72:CALCulate:DELTamarker<n>:Y ......................................... 72:CALCulate:DEMod:AM:RESult:CURRent?..................... 72:CALCulate:DEMod: AM:RESult:MINimum?.................... 73
49

GSP-930 User Manual:CALCulate:DEMod: AM:RESult:MAXimum?...................73:CALCulate:DEMod:AM:STATe............................................74:CALCulate:DEMod:EARPhone:STATe.............................74:CALCulate:DEMod:FM:RESult:CURRent? ......................75:CALCulate:DEMod: FM:RESult:MINimum? .....................75:CALCulate:DEMod: FM:RESult:MAXimum? ...................76:CALCulate:DEMod:FM:STATe ............................................76:CALCulate:JITTer:STATe ....................................................77:CALCulate:JITTer:CARRier:POWer? ...............................77:CALCulate:JITTer: PHASe? ................................................77:CALCulate:JITTer:TIME?......................................................78:CALCulate:LIMit<n>:CLEar..................................................78:CALCulate:LIMit<n>:DATA ..................................................78:CALCulate:LIMit:FAIL? ..........................................................78:CALCulate:LIMit:LOW ............................................................79:CALCulate:LIMit:HIGH ...........................................................79:CALCulate:LIMit<n>:MARKer..............................................79:CALCulate:LIMit:PASS:MODE............................................80:CALCulate:LIMit:STATe........................................................80:CALCulate:LIMit<n>:TRACe................................................80:CALCulate:MARKer:AOFF ...................................................81:CALCulate:MARKer<n>:FCOunt:RESolution.................81:CALCulate:MARKer<n>:FCOunt:RESolution:AUTO...81:CALCulate:MARKer<n>:FCOunt:STATe.........................82:CALCulate:MARKer<n>:FCOunt:X?..................................82:CALCulate:MARKer<n>:NOISe:STATe............................83:CALCulate:MARKer<n>:NOISe:Y?.....................................83:CALCulate:MARKer<n>:PEAK ............................................83:CALCulate:MARKer<n>:PEAK:CTRack:STATe............84:CALCulate:MARKer:PEAK:DATA? ....................................84:CALCulate:MARKer<n>:PEAK:EXCursion......................84:CALCulate:MARKer:PEAK:SORT:TYPE..........................85:CALCulate:MARKer:PEAK:TABLe:STATe .....................85:CALCulate:MARKer:PEAK:THReshold ............................86:CALCulate:MARKer:PEAK:THReshold:STATe .............86:CALCulate:MARKer<n>:SET ...............................................86:CALCulate:MARKer<n>:STATe..........................................87:CALCulate:MARKer:TABLe:STATe..................................8750

エラー! スタイルが定義されていません。
:CALCulate:MARKer<n>:TRACe......................................... 88:CALCulate:MARKer<n>:TRACe:AUTO........................... 88:CALCulate:MARKer<n>:TYPE............................................ 89:CALCulate:MARKer<n>:X .................................................... 89:CALCulate:MARKer<n>:Y? .................................................. 89:CALCulate:MATH:PDIF......................................................... 90:CALCulate:MATH:LDIF ......................................................... 90:CALCulate:MATH:LOFF ....................................................... 90:CALCulate:NDB:STATe........................................................ 91:CALCulate:NDB:BANDwidth|BWIDth? ............................ 91:CALCulate:NORMalize:STATe........................................... 91:CALCulate:OCBW:STATe.................................................... 92:CALCulate:OCBW:BANDwidth|BWIDth?........................ 92:CALCulate:OCBW:CHPower?............................................. 92:CALCulate:OCBW:POWer?.................................................. 93:CALCulate:PMETer:POWer?............................................... 93:CALCulate:PMETer:LIMit:STATe...................................... 93:CALCulate:PMETer:LIMit:FAIL?........................................ 94:CALCulate:SEM:STATe........................................................ 94:CALCulate:SEM:OFFSet<n>:RESult?.............................. 94:CALCulate:TOI:DIFFerential?............................................. 95:CALCulate:TOI:FREQuency:STEPsize?......................... 95:CALCulate:TOI:LIMit:STATe .............................................. 95:CALCulate:TOI:RESult?........................................................ 96:CALCulate:TOI:STATe.......................................................... 96

CONFigure
Commands:CONFigure:MODE ................................................................... 97

DISPlay
Commands:DISPlay:BRIGhtness .............................................................. 98:DISPlay:ENABle ....................................................................... 98:DISPlay:DEMod[:WINDow]:TRACe:X[:SCALe]:AUTO 99:DISPlay:DEMod[:WINDow]:TRACe:X[:SCALe]:PDIVision ................................................................. 99:DISPlay:DEMod[:WINDow]:TRACe:X[:SCALe]:RPOSition.............................................................. 100
51

GSP-930 User Manual:DISPlay:DEMod[:WINDow]:TRACe:X[:SCALe]:RVALue....................................................................................................... 100:DISPlay:DEMod[:WINDow]:TRACe:Y[:SCALe]:AUTO....................................................................................................... 101:DISPlay:DEMod[:WINDow]:TRACe:Y[:SCALe]:PDIVision ............................................................... 101:DISPlay:DEMod[:WINDow]:TRACe:Y[:SCALe]:RPOSition.............................................................. 101:DISPlay:DEMod[:WINDow]:TRACe:Y[:SCALe]:RVALue....................................................................................................... 102:DISPlay[:WINDow]:NORMal............................................... 102:DISPlay[:WINDow]:SPECtrogram.................................... 102:DISPlay[:WINDow]:SPLit:NORMal .................................. 103:DISPlay[:WINDow]:SPLit:NORMal:ALTernate............ 103:DISPlay[:WINDow]:SPLit:NORMal:TYPE...................... 103:DISPlay[:WINDow]:SPLit:SPECtrogram ....................... 103:DISPlay[:WINDow]:SPLit:TOPO ...................................... 104:DISPlay[:WINDow]:SPLit:TOPO ...................................... 104:DISPlay[:WINDow]:TRACe<n>:MODE ........................... 104:DISPlay[:WINDow]:TRACe:Y:DLINe ............................... 104:DISPlay[:WINDow]:TRACe:Y:DLINe:STATe................ 105:DISPlay[:WINDow]:TRACe:Y[:SCALe]:NRLevel........ 105:DISPlay[:WINDow]:TRACe:Y[:SCALe]:NRPosition .. 106:DISPlay[:WINDow]:TRACe:Y[:SCALe]:PDIVision ..... 106:DISPlay[:WINDow]:TRACe:Y[:SCALe]:POSition....... 106:DISPlay[:WINDow]:TRACe:Y[:SCALe]:RLEVel.......... 107:DISPlay[:WINDow]:TRACe:Y[:SCALe]:RLEVel:OFFSet....................................................................................................... 107:DISPlay[:WINDow]:TRACe:Y[:SCALe]:SPACing....... 108:DISPlay[:WINDow]:TRACe:Y[:SCALe]:STATe........... 108

FORMatCommands:FORMat:TRACe:DATA........................................................ 109

INITiate
Commands:INITiate:CONTinuous........................................................... 110:INITiate[:IMMediate] ............................................................ 110

52

エラー! スタイルが定義されていません。
INPut Commands :INPut:ATTenuation............................................................... 111:INPut:ATTenuation:AUTO ................................................. 111:INPut:IMPedance................................................................... 112:INPut:OFFSet ......................................................................... 112

MMEMorycommands:MMEMory:CATalog?............................................................. 113:MMEMory:CDIRectory......................................................... 113:MMEMory:COPY.................................................................... 114:MMEMory:DELete ................................................................. 114:MMEMory:DESTination ....................................................... 114:MMEMory:LOAD:CORRection.......................................... 115:MMEMory:LOAD:LIMit......................................................... 115:MMEMory:LOAD:PMET....................................................... 115:MMEMory:LOAD:SEQuence.............................................. 115:MMEMory:LOAD:STATe..................................................... 116:MMEMory:LOAD:TRACe .................................................... 116:MMEMory:MOVE.................................................................... 116:MMEMory:REName ............................................................... 117:MMEMory:STORe:CORRection ....................................... 117:MMEMory:STORe:LIMit....................................................... 117:MMEMory:STORe:PMET .................................................... 117:MMEMory:STORe:SCReen ................................................ 118:MMEMory:STORe:SEQuence ........................................... 118:MMEMory:STORe:STATe .................................................. 118:MMEMory:STORe:TRACe.................................................. 119

OUTPutcommands:OUTPut[:STATe]................................................................... 119

SENse commands [:SENSe]:ACPR:ACHannel<n>:BANDwidth|BWIDth . 123[:SENSe]:ACPR:ACHannel<n>:HLIMit............................ 123[:SENSe]:ACPR:ACHannel<n>:LLIMit ............................ 124[:SENSe]:ACPR:ACHannel<n>:OFFSet ......................... 124[:SENSe]:ACPR:BANDwidth|BWIDth.............................. 124[:SENSe]:ACPR:HLIMit ........................................................ 125
53

GSP-930 User Manual[:SENSe]:ACPR:LLIMit......................................................... 125[:SENSe]:ACPR:HELP.......................................................... 125[:SENSe]:ACPR:SPACe....................................................... 126[:SENSe]:ASET:AMPLitude................................................ 126[:SENSe]:ASET:AMPLitude:AUTO .................................. 126[:SENSe]:ASET:RUN............................................................. 127[:SENSe]:ASET:SPAN.......................................................... 127[:SENSe]:ASET:SPAN:AUTO ............................................ 127[:SENSe]:AVERage:COUNt ................................................ 128[:SENSe]:AVERage:STATe................................................. 128[:SENSe]:AVERage:TYPE ................................................... 128[:SENSe]:BANDwidth|BWIDth[:RESolution] ................ 129[:SENSe]:BANDwidth|BWIDth[:RESolution]:AUTO... 129[:SENSe]:BANDwidth|BWIDth:VIDeo.............................. 129[:SENSe]:BANDwidth|BWIDth:VIDeo:AUTO ................ 130[:SENSe]:CHANnel:SPACe:DOWN .................................. 130[:SENSe]:CHANnel:SPACe:UP ......................................... 130[:SENSe]:CNR:CHANnel:SPACe ...................................... 131[:SENSe]:CNR:DELTamarker:MODE............................... 131[:SENSe]:CORRection:CSET<n>:DATA ........................ 131[:SENSe]:CORRection:CSET<n>:STATe...................... 132[:SENSe]:CORRection:CSET<n>:DELete ..................... 132[:SENSe]:CSO:CHANnel:SPACe...................................... 132[:SENSe]:CTB:CHANnel:SPACe ...................................... 133[:SENSe]:DEMod:EARPone:TYPE.................................... 133[:SENSe]:DEMod:EARPone:VOLume ............................. 133[:SENSe]:DEMod:EARPone:GAIN .................................... 134[:SENSe]:DEMod:FILTer:LPASs....................................... 134[:SENSe]:DEMod:IFBW......................................................... 135[:SENSe]:DETector[:FUNCtion] ....................................... 135[:SENSe]:DETector[:FUNCtion]:AUTO.......................... 135[:SENSe]:EMIFilter:STATe ................................................. 136[:SENSe]:EMIFilter:BANDwidth|BWIDth[:RESolution]....................................................................................................... 136[:SENSe]:FREQuency:CENter........................................... 137[:SENSe]:FREQuency:CENter:STEP .............................. 137[:SENSe]:FREQuency:CENter:STEP:AUTO................. 137[:SENSe]:FREQuency:OFFSet.......................................... 138
54

エラー! スタイルが定義されていません。
[:SENSe]:FREQuency:SPAN.............................................. 138[:SENSe]:FREQuency:SPAN:FULL.................................. 138[:SENSe]:FREQuency:SPAN:PREVious......................... 138[:SENSe]:FREQuency:STARt ............................................ 139[:SENSe]:FREQuency:STOP.............................................. 139[:SENSe]:LIMit<n>:DELete ................................................. 139[:SENSe]:JITTer:OFFSet:STARt...................................... 140[:SENSe]:JITTer:OFFSet:STOP ....................................... 140[:SENSe]:NDB:BANDwidth|BWIDth................................. 140[:SENSe]:OCBW:BANDwidth|BWIDth............................. 140[:SENSe]:OCBW:PERCent.................................................. 141[:SENSe]:OCBW:SPACe...................................................... 141[:SENSe]:OCBW:SPAN......................................................... 141[:SENSe]:PMETer:FREQuency.......................................... 142[:SENSe]:PMETer:HLIMit .................................................... 142[:SENSe]:PMETer:HOLD:STATe ...................................... 142[:SENSe]:PMETer:LLIMit..................................................... 143[:SENSe]:PMETer:PSENsor:MODE ................................. 143[:SENSe]:PMETer:RECording:TIME................................. 143[:SENSe]:PMETer:RECording:TIME:STEP .................... 144[:SENSe]:POWer[:RF]:GAIN ............................................... 144[:SENSe]:SEMask:BANDwidth|BWIDth:INTegration.. 144[:SENSe]:SEMask:BANDwidth|BWIDth[:RESolution]145[:SENSe]:SEMask:BANDwidth|BWIDth[:RESolution]:AU
TO ................................................................................................ 145[:SENSe]:SEMask:CARRier:AUTO................................... 146[:SENSe]:SEMask:CARRier:CPSD................................... 146[:SENSe]:SEMask:CARRier: POWer................................ 147[:SENSe]:SEMask:FREQuency:SPAN............................. 147[:SENSe]:SEMask:GWLan:MODulation........................... 147[:SENSe]:SEMask:HELP:STATe ....................................... 148[:SENSe]:SEMask:NWLan:CHANnel:
BANDwidth|BWIDth............................................................... 148[:SENSe]:SEMask:OFFSet<n>:ADDition:
BANDwidth|BWIDth[:RESolution]?.................................. 148[:SENSe]:SEMask:OFFSet<n>:ADDition:
FREQuency:STARt?.............................................................. 149[:SENSe]:SEMask:OFFSet<n>:ADDition:
55

GSP-930 User Manual
FREQuency:STOP?............................................................... 149[:SENSe]:SEMask:OFFSet<n>:ADDition:STARt:ABSolute? .................................................................. 150[:SENSe]:SEMask:OFFSet<n>:ADDition:STARt:
RELative?.................................................................................. 150[:SENSe]:SEMask:OFFSet<n>:ADDition:STATe?...... 150[:SENSe]:SEMask:OFFSet<n>:ADDition:
STOP:ABSolute?.................................................................... 151[:SENSe]:SEMask:OFFSet<n>:ADDition:STOP:
RELative?.................................................................................. 151[:SENSe]:SEMask:OFFSet<n>:BANDwidth|
BWIDth[:RESolution] ............................................................ 152[:SENSe]:SEMask:OFFSet<n>:BANDwidth|BWIDth[:RE
Solution]:AUTO....................................................................... 152[:SENSe]:SEMask:OFFSet<n>:FREQuency:STARt... 153[:SENSe]:SEMask:OFFSet<n>:FREQuency:STOP.... 153[:SENSe]:SEMask:OFFSet<n>:STARt:ABSolute ....... 153[:SENSe]:SEMask:OFFSet<n>:STARt:RELative ........ 154[:SENSe]:SEMask:OFFSet<n>:STATe ........................... 154[:SENSe]:SEMask:OFFSet<n>:STOP:ABSolute......... 154[:SENSe]:SEMask:OFFSet<n>:STOP:ABSolute:COUPle....................................................................................................... 155[:SENSe]:SEMask:OFFSet<n>:STOP:RELative.......... 155[:SENSe]:SEMask:OFFSet<n>:STOP:RELative:COUPle....................................................................................................... 156[:SENSe]:SEMask:OFFSet<n>:TEST.............................. 156[:SENSe]:SEMask:SELect................................................... 157[:SENSe]:SEMask:TYPE ...................................................... 157[:SENSe]:SEMask:W3GPP:DUPLex:TYPE.................... 157[:SENSe]:SEMask:W3GPP:FDD:ADDition:Limit.......... 158[:SENSe]:SEMask:W3GPP:FDD:ADDition:MOPower 158[:SENSe]:SEMask:W3GPP:FDD:MOPower ................... 159[:SENSe]:SEMask:W3GPP:FDD:TRANsmit:MODE .... 159[:SENSe]:SEMask:W3GPP:TDD:CHIP:RATE................ 159[:SENSe]:SEMask:W3GPP:TDD:MOPower ................... 160[:SENSe]:SEMask:W3GPP:TDD:TRANsmit:MODE .... 160[:SENSe]:SEMask:WIMax:CHANnel:BANDwidth|BWIDth....................................................................................................... 160
56

エラー! スタイルが定義されていません。
[:SENSe]:SEQuence<n>:DELete ...................................... 161[:SENSe]:SWEep:EGATe:DELay ....................................... 161[:SENSe]:SWEep:EGATe:LENGth .................................... 161[:SENSe]:SWEep:EGATe:STATe ...................................... 162[:SENSe]:SWEep:TIME ......................................................... 162[:SENSe]:SWEep:TIME:AUTO............................................ 162[:SENSe]:TOI:REFerence .................................................... 163[:SENSe]:TOI:LIMit ................................................................ 163

SEQuencecommands:SEQuence:MODE .................................................................. 164:SEQuence:TEST:ACTive.................................................... 164

SOURcecommands

SOURce Commands ............................................................. 164:SOURce:POWer[:LEVel][:IMMediate][:AMPLitude]. 165:SOURce:POWer[:LEVel][:IMMediate][:AMPLitude]:OFFSet...................................................................................... 165:SOURce:POWer[:LEVel][:IMMediate][:AMPLitude]:STEP .......................................................................................... 165:SOURce:POWer[:LEVel][:IMMediate][:AMPLitude]:STEP:AUTO............................................................................. 166:SOURce:POWer:MODE....................................................... 166:SOURce:POWer:SWEep...................................................... 166

SYSTemcommands:SYSTem:ALARm:STATe .................................................... 168:SYSTem:CLOCk<n>:DATE................................................ 169:SYSTem:CLOCk<n>:MODE............................................... 169:SYSTem:CLOCk<n>:STATe.............................................. 170:SYSTem:CLOCk<n>:TIME ................................................. 170:SYSTem:COMMunicate:GPIB[:SELF]:ADDRess....... 171:SYSTem:COMMunicate:LAN:ADDRess........................ 171:SYSTem:COMMunicate:LAN:MASK............................... 171:SYSTem:COMMunicate:LAN:GATEway ....................... 171:SYSTem:COMMunicate:LAN:CONFigure..................... 172:SYSTem:COMMunicate:LCI .............................................. 172:SYSTem:COMMunicate:SERial[:RECeive]:BAUD..... 172
57

GSP-930 User Manual:SYSTem:COMMunicate:USB:MODE.............................. 172:SYSTem:DATE....................................................................... 173:SYSTem:ERRor:CLEar........................................................ 173:SYSTem:ERRor[:NEXT]? ................................................... 173:SYSTem:KLOCk .................................................................... 173:SYSTem:PRESet................................................................... 174:SYSTem:PRESet:TYPE ...................................................... 174:SYSTem:PRESet:USER:SAVE......................................... 174:SYSTem:REBoot ................................................................... 174:SYSTem:SHUTdown ............................................................ 174:SYSTem:TIME ........................................................................ 175:SYSTem:UPDate................................................................... 175:SYSTem:VERSion:HARDware?........................................ 175:SYSTem:VERSion:SOFTware?........................................ 175

STATuscommands

58:STATus:OPERation:CONDition?..................................... 177:STATus:OPERation:ENABle............................................. 170:STATus:OPERation[:EVENt]?.......................................... 170:STATus:OPERation:NTRansition.................................... 170:STATus:OPERation:PTRansition.................................... 171:STATus:QUEStionable:CONDition?............................... 171:STATus:QUEStionable:ENABle ....................................... 172:STATus:QUEStionable[:EVENt]?.................................... 172:STATus:QUEStionable:NTRansition.............................. 173:STATus:QUEStionable:PTRansition.............................. 173:STATus:QUEStionable:FREQuency:CONDition?...... 174:STATus:QUEStionable:FREQuency:ENABle.............. 174:STATus:QUEStionable:FREQuency[:EVENt]?........... 174:STATus:QUEStionable:FREQuency:NTRansition..... 175:STATus:QUEStionable:FREQuency:PTRansition..... 175:STATus:QUEStionable:ACPLimit:CONDition? .......... 176:STATus:QUEStionable:ACPLimit:ENABle................... 176:STATus:QUEStionable:ACPLimit[:EVENt]? ............... 177:STATus:QUEStionable:ACPLimit:NTRansition ......... 177:STATus:QUEStionable:ACPLimit:PTRansition.......... 178:STATus:PRESet.................................................................... 178

エラー! スタイルが定義されていません。
TRACe
commands:TRACe[:DATA]? .................................................................... 179

TRIGgercommands:TRIGger[:SEQuence]:DELay............................................. 180:TRIGger[:SEQuence]:DEMod:DELay ............................. 180:TRIGger[:SEQuence]:DEMod:LEVel............................... 181:TRIGger[:SEQuence]:DEMod:MODE.............................. 181:TRIGger[:SEQuence]:DEMod:SLOPe ............................ 181:TRIGger[:SEQuence]:DEMod:SOURce ......................... 182:TRIGger[:SEQuence]:EXTernal:SLOPe........................ 182:TRIGger[:SEQuence]:MODE ............................................. 182:TRIGger[:SEQuence]:PMETer:SOURce ....................... 183:TRIGger[:SEQuence]:SOURce......................................... 183:TRIGger[:SEQuence]:VIDeo:FREQuency..................... 183:TRIGger[:SEQuence]:VIDeo:LEVel................................. 184:TRIGger[:SEQuence]:VIDeo:SLOPe .............................. 184

UNIT commands :UNIT:PMETer:POWer........................................................... 185:UNIT:POWer ............................................................................ 185

59

GSP-930 User Manual

SCPI Commands*CLS ..............................................................................................60*IDN................................................................................................60*ESE...............................................................................................61*ESR? ............................................................................................61*OPC .............................................................................................61*RST ..............................................................................................62*SRE ..............................................................................................62*STB?............................................................................................62*TST ..............................................................................................62*WAI ...............................................................................................63

S et

*CLS
Description

The *CLS command clears the Standard Event
Status, Operation Status and Questionable Status
registers. The corresponding Enable registers in eachof the above registers are not cleared.
If a <NL> newline code immediately precedes a *CLS
command, the Error Que and the MAV bit in the
Status Byte Register is also cleared.

Syntax

*CLS
Query

*IDN
Description

Queries the manufacturer, model number, serial
number, and firmware version of the instrument.

Query Syntax

*IDN?

60

エラー! スタイルが定義されていません。
Return parameter <string> Returns the instrument identification as a stringin the following format:
GW-INSTEK,GSP-930, XXXXXXXXXXXX,V.X.X.X.X
Manufacturer: GW-INSTEKModel number : GSP-930
Serial number : XXXXXXXXXXXX
Firmware version : V.X.X.X.X

S et
Query

*ESE
Description

Sets or queries the Standard Event Status Enableregister.

Syntax*ESE <NR1>

Query Syntax*ESE?
Parameter<NR1>
Return parameter <NR1>

Query

*ESR?
Description

0~255
Returns the bit sum of the Standard Event
Status Enable register.

Queries the Standard Event Status register. The
Event Status register is cleared after it is read.

Query Syntax*ESR?
Return parameter <NR1>

Returns the bit sum of the Standard Event
Status register and clears the register.
S et

*OPC
Description

Query

The *OPC command sets the OPC bit (bit0) of theStandard Event Status Register when all current
commands have been processed.
The *OPC? Query returns 1 when all the outstanding
commands have completed.

61

GSP-930 User Manual
Syntax*OPC

Query Syntax*OPC?
Return parameter 1

Returns 1 when all the outstanding commands
have completed.
S et

*RST
Description

Performs a device reset. Configures the unit to a
known configuration (default settings). This known
configuration is independent of the usage history.

Syntax

*RST
S et
Query

*SRE
Description

Sets or queries the Service Request Enable register.
The Service Request Enable register determines
which registers of the Status Byte register are able togenerate service requests.

Syntax*SRE <NR1>

Query Syntax*SRE?
Parameter<NR1>
Return parameter <NR1>

Query

*STB?
Description

Queries the bit sum of the Status Byte register withMSS (Master summary Status).

Query Syntax*STB?
Return parameter <NR1>

*TST

62

0~255
Returns the bit sum of the Service Request
Enable register.

Returns the bit sum of the Status Byte registerwith the MSS bit (bit 6).
Queryエラー! スタイルが定義されていません。
Description

Executes a self test.

Query Syntax*TST?
Return parameter 0<NR1>

Returns “0” if there are no errors.
Returns an error code <NR1> if there is an error.
S et

*WAI
Description

Prevents any other commands or queries from being
executed until all outstanding commands havecompleted.

Syntax*WAI

CALCulate Commands:CALCulate:ACPR:ACHannel<n>:HLIMit:FAIL? ............ 65:CALCulate:ACPR:ACHannel<n>:LLIMit:FAIL?............. 66:CALCulate:ACPR:ACHannel<n>:LOWer?....................... 66:CALCulate:ACPR:ACHannel<n>:UPPer? ....................... 66:CALCulate:ACPR:ACHannel<n>:STATe ........................ 67:CALCulate:ACPR:CHANnel:HLIMit:FAIL? ..................... 67:CALCulate:ACPR:CHAN:LLIMit:FAIL?............................ 68:CALCulate:ACPR:CHPower?.............................................. 68:CALCulate:ACPR:STATe..................................................... 68:CALCulate:CNR:RESult?...................................................... 69:CALCulate:CNR:STATe........................................................ 69:CALCulate:CSO:RESult?...................................................... 69:CALCulate:CSO:STATe........................................................ 70:CALCulate:CTB:RESult? ...................................................... 70:CALCulate:CTB:STATe ........................................................ 70:CALCulate:CTB:RESTart..................................................... 71:CALCulate:DELTamarker<n>:PAIR:SPAN..................... 71:CALCulate:DELTamarker<n>:PAIR:CENTer................. 71:CALCulate:DELTamarker<n>:X.......................................... 72:CALCulate:DELTamarker<n>:Y.......................................... 72:CALCulate:DEMod:AM:RESult:CURRent? ..................... 72:CALCulate:DEMod: AM:RESult:MINimum? .................... 73
63

GSP-930 User Manual:CALCulate:DEMod: AM:RESult:MAXimum? ...................73:CALCulate:DEMod:AM:STATe............................................74:CALCulate:DEMod:EARPhone:STATe.............................74:CALCulate:DEMod:FM:RESult:CURRent? ......................75:CALCulate:DEMod: FM:RESult:MINimum? .....................75:CALCulate:DEMod: FM:RESult:MAXimum?....................76:CALCulate:DEMod:FM:STATe ............................................76:CALCulate:JITTer:STATe ....................................................77:CALCulate:JITTer:CARRier:POWer? ...............................77:CALCulate:JITTer: PHASe? ................................................77:CALCulate:JITTer:TIME?......................................................78:CALCulate:LIMit<n>:CLEar ..................................................78:CALCulate:LIMit<n>:DATA...................................................78:CALCulate:LIMit:FAIL? ..........................................................78:CALCulate:LIMit:LOW ............................................................79:CALCulate:LIMit:HIGH ...........................................................79:CALCulate:LIMit<n>:MARKer..............................................79:CALCulate:LIMit:PASS:MODE ............................................80:CALCulate:LIMit:STATe........................................................80:CALCulate:LIMit<n>:TRACe ................................................80:CALCulate:MARKer:AOFF....................................................81:CALCulate:MARKer<n>:FCOunt:RESolution.................81:CALCulate:MARKer<n>:FCOunt:RESolution:AUTO ...81:CALCulate:MARKer<n>:FCOunt:STATe.........................82:CALCulate:MARKer<n>:FCOunt:X?..................................82:CALCulate:MARKer<n>:NOISe:STATe............................83:CALCulate:MARKer<n>:NOISe:Y?.....................................83:CALCulate:MARKer<n>:PEAK ............................................83:CALCulate:MARKer<n>:PEAK:CTRack:STATe............84:CALCulate:MARKer:PEAK:DATA? ....................................84:CALCulate:MARKer<n>:PEAK:EXCursion......................84:CALCulate:MARKer:PEAK:SORT:TYPE..........................85:CALCulate:MARKer:PEAK:TABLe:STATe .....................85:CALCulate:MARKer:PEAK:THReshold ............................86:CALCulate:MARKer:PEAK:THReshold:STATe .............86:CALCulate:MARKer<n>:SET................................................86:CALCulate:MARKer<n>:STATe..........................................87:CALCulate:MARKer:TABLe:STATe ..................................8764

エラー! スタイルが定義されていません。
:CALCulate:MARKer<n>:TRACe......................................... 88:CALCulate:MARKer<n>:TRACe:AUTO ........................... 88:CALCulate:MARKer<n>:TYPE............................................ 89:CALCulate:MARKer<n>:X .................................................... 89:CALCulate:MARKer<n>:Y? .................................................. 89:CALCulate:MATH:PDIF......................................................... 90:CALCulate:MATH:LDIF ......................................................... 90:CALCulate:MATH:LOFF ....................................................... 90:CALCulate:NDB:STATe........................................................ 91:CALCulate:NDB:BANDwidth|BWIDth? ............................ 91:CALCulate:NORMalize:STATe........................................... 91:CALCulate:OCBW:STATe.................................................... 92:CALCulate:OCBW:BANDwidth|BWIDth? ........................ 92:CALCulate:OCBW:CHPower?............................................. 92:CALCulate:OCBW:POWer?.................................................. 93:CALCulate:PMETer:POWer?............................................... 93:CALCulate:PMETer:LIMit:STATe...................................... 93:CALCulate:PMETer:LIMit:FAIL? ........................................ 94:CALCulate:SEM:STATe ........................................................ 94:CALCulate:SEM:OFFSet<n>:RESult?.............................. 94:CALCulate:TOI:DIFFerential?............................................. 95:CALCulate:TOI:FREQuency:STEPsize? ......................... 95:CALCulate:TOI:LIMit:STATe............................................... 95:CALCulate:TOI:RESult?........................................................ 96:CALCulate:TOI:STATe.......................................................... 96:CALCulate:ACPR:ACHannel<n>:HLIMit:
FAIL?

Query

Description

Returns the ACPR upper limit pass/fail judgment forthe selected channel.

Query Syntax:CALCulate:ACPR:ACHannel<n>:HLIMit:FAIL?

Parameter<n>Return parameter 0
1<NR1>adjacent channel 1~3<boolean>Pass<boolean>Fail65

GSP-930 User Manual
Query Example:CALC:ACPR:ACH<n>:HLIM:FAIL?
>0:CALCulate:ACPR:ACHannel<n>:LLIMit:
FAIL?

Query

Description

Returns the ACPR lower limit pass/fail judgment forthe selected channel.

Query Syntax:CALCulate:ACPR:ACHannel<n>:LLIMit:FAIL?

Parameter<n>Return parameter 0
1
Query Example<NR1>adjacent channel 1~3<boolean>Pass<boolean>Fail:CALC:ACPR:ACH1:LLIM:FAIL?
>0:CALCulate:ACPR:ACHannel<n>:LOWer?

Query

Description

Returns the ACPR adjacent channel lower bandwidth
for the selected channel.

Query Syntax:CALCulate:ACPR:ACHannel<n>:LOWer?

Parameter<n>Return parameter <NR3>
Query Example<NR1>adjacent channel 1~3
Adjacent bandwidth in dBm:CALC:ACPR:ACH1:LOW?
>1.801e+01:CALCulate:ACPR:ACHannel<n>:UPPer?

Query

Description

Returns the ACPR adjacent channel upper bandwidth
for the selected channel.

Query Syntax:CALCulate:ACPR:ACHannel<n>:UPPer?

66

エラー! スタイルが定義されていません。
Parameter<n>Return parameter <NR3>
Query Example<NR1>adjacent channel 1~3
Adjacent bandwidth in dBm:CALC:ACPR:ACH1:UPP?
>1.921e+01
S et:CALCulate:ACPR:ACHannel<n>:STATe

Query

Description

Sets or queries the state of the selected adjacentchannel.

Syntax:CALCulate:ACPR:ACHannel<n>:STATe {OFF|ON|0|1}

Query Syntax:CALCulate:ACPR:ACHannel<n>:STATe?

Parameter

<n>
0
1
OFF
ON

Return parameter 0
1
Query Example<NR1>adjacent channel 1~3
Disable the selected channel.
Enable the selected channel.
Disable the selected channel.
Enable the selected channel.
The selected channel is enabled.
The selected channel is disabled.

:CALC:ACPR:ACH1:STAT?
>1:CALCulate:ACPR:CHANnel:HLIMit:FAIL?

Query

Description

Returns the ACPR upper limit pass/fail judgment forthe main channel.

Query Syntax:CALCulate:ACPR:CHANnel:HLIMit:FAIL?

Return parameter 0
1
Query Example<boolean>Pass<boolean>Fail:CALC:ACPR:CHAN:HLIM:FAIL?
>067

GSP-930 User Manual

Query:CALCulate:ACPR:CHAN:LLIMit:FAIL?
Description

Returns the ACPR lower limit pass/fail judgment forthe main channel.

Query Syntax:CALCulate:ACPR:CHAN:LLIMit:FAIL?

Return parameter 0
1
Query Example<boolean>Pass<boolean>Fail:CALC:ACPR:CHAN:LLIM:FAIL?
>0
Query

:CALCulate:ACPR:CHPower?
Description

Returns the ACPR main channel power in the current
chosen unit (default dBm).

Query Syntax:CALCulate:ACPR:CHPower?

Return parameter <NR3>
Query Example

Power:CALC:ACPR:CHP?
>-1.028e+02
S et
Query

:CALCulate:ACPR:STATe
Description

Sets or queries the state of the main channel.

Syntax:CALCulate:ACPR:STATe {OFF|ON|0|1}

Query Syntax:CALCulate:ACPR:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1

68Main channel is disabled.
Main channel is enabled.
Main channel is disabled.
Main channel is enabled.
Main channel is enabled.
Main channel is disabled.

エラー! スタイルが定義されていません。
Query Example:CALC:ACPR:STAT?
>1
Query

:CALCulate:CNR:RESult?
Description

Returns the CNR measurement result in dB.

Query Syntax:CALCulate:CNR:RESult?

Return parameter <NR3>
Query Example

CNR measurement in dB:CALC:CNR:RES?
>-4.959e+01
S et
Query

:CALCulate:CNR:STATe
Description

Sets or queries the state of the CNR measurement
function.

Syntax:CALCulate:CNR:STATe {ON|OFF|1|0}

Query Syntax:CALCulate:CNR:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1
Query Example

CNR is off.
CNR is on.
CNR is off.
CNR is on.
CNR is off.
CNR is on.

:CALC:CNR:STAT?
>1:CALCulate:CSO:RESult?

Query

Description

Returns the CSO measurement result in dB.

Query Syntax:CALCulate:CSO:RESult?

69

GSP-930 User Manual
Return parameter <NR3>
Query Example

CSO measurement in dB:CALC:CSO:RES?
>4.04e+00
S et
Query

:CALCulate:CSO:STATe
Description

Sets or queries the state of the CSO measurement
function.

Syntax:CALCulate:CSO:STATe {ON|OFF|1|0}

Query Syntax:CALCulate:CSO:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1
Query Example

CSO is off.
CSO is on.
CSO is off.
CSO is on.
CSO is off.
CSO is on.

:CALC:CSO:STAT?
>1
Query

:CALCulate:CTB:RESult?
Description

Returns the CTB measurement result in dB.

Query Syntax:CALCulate:CTB:RESult?

Return parameter <NR3>
Query Example

CTB measurement in dB:CALC:CTB:RES?
>-4.237e+01
S et:CALCulate:CTB:STATe

Query

Description

Sets or queries the state of the CTB measurement
function.

Syntax:CALCulate:CTB:STATe {ON|OFF|1|0}70

エラー! スタイルが定義されていません。
Query Syntax:CALCulate:CTB:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1
Query Example

CTB is off.
CTB is on.
CTB is off.
CTB is on.
CTB is off.
CTB is on.

:CALC:CTB:STAT?
>0
S et:CALCulate:CTB:RESTart
Description

Restarts the CTB measurement.

Syntax:CALCulate:CTB:RESTart:CALCulate:DELTamarker<n>:PAIR:SPAN

S et

Description

Sets the span between the chosen marker and thedelta marker.

Syntax:CALCulate:DELTamarker<n>:PAIR:SPAN <freq>

Parameter<freq>

Example:CALC:DELT1:PAIR:SPAN 1e+9

<NR3> frequency of span in Hz.

:CALCulate:DELTamarker<n>:PAIR:CENTer

S etDescription

Sets the frequency span between the chosen marker
and the delta marker, centered from the center
frequency.

Syntax:CALCulate:DELTamarker<n>:PAIR:CENTer <freq>

Parameter<freq>

Example:CALC:DELT1:PAIR:CENT 1e+9

<NR3> frequency of span in Hz.

71

GSP-930 User Manual
S et
Query

:CALCulate:DELTamarker<n>:X
Description

Sets or queries the selected delta marker position.

Syntax:CALCulate:DELTamarker<n>:X <freq>

Query Syntax:CALCulate:DELTamarker<n>:X?

Parameter<n>Return parameter <freq>
ExampleMarker number.
<NR3> frequency in Hz.

:CALC:DELT1:X?
>1e+9
S et

:CALCulate:DELTamarker<n>:Y

Query

Description

Returns the selected delta marker Y axis value. The Y
axis units are the same as the units used for the
trace (default=dBm).

Query Syntax:CALCulate:DELTamarker<n>:Y?

Parameter<n>Return parameter <freq>
ExampleMarker number.
<NR3> frequency in Hz.

:CALC:DELT1:Y?
>-1.032e+1:CALCulate:DEMod:AM:RESult:CURRent?

Query

Description

Returns the measurement results for AM
demodulation as a comma separated string.

Query Syntax:CALCulate:DEMod:AM:RESult:CURRent?

Return parameter <depth,mode,rate,power,carr freq, offset,sinad>
depth
Modulation depth in %.
mode
AM.
rate
Modulation rate in Hz.
72エラー! スタイルが定義されていません。
power
carr freq.
offsetsinad
Query Example

Carrier power in dBm.
Carrier frequency in Hz.
Carrier frequency offset in Hz.
Signal to noise and distortion ratio in dB:CALC:DEM:AM:RES:CURR?
>9.840e+1,1.02e+2,-1.12e+1,3.21e+1,-1.14e+1,
2.7e+3,1.61e+1:CALCulate:DEMod: AM:RESult:MINimum?

Query

Description

Returns the minimum recorded measurement resultsfor AM demodulation as a comma separated string.

Query Syntax:CALCulate:DEMod:AM:RESult:MINimum?

Return parameter <depth,mode,rate,power,carr freq, offset,sinad>
depth
Modulation depth in %.
mode
AM.
rate
Modulation rate in Hz.
power
Carrier power in dBm.
carr freq.
Carrier frequency in Hz.
offsetCarrier frequency offset in Hz.
sinad
Signal to noise and distortion ratio in dB
Query Example:CALC:DEM:AM:RES:MIN?
>9.840e+1,1.02e+2,-1.12e+1,3.21e+1,-1.14e+1,
2.7e+3,1.61e+1:CALCulate:DEMod: AM:RESult:MAXimum?

Query

Description

Returns the maximum recorded measurement resultsfor AM demodulation as a comma separated string.

Query Syntax:CALCulate:DEMod:AM:RESult:MAXimum?

Return parameter <depth,mode,rate,power,carr freq, offset,sinad>
depth
Modulation depth in %.
mode
AM.
rate
Modulation rate in Hz.
73

GSP-930 User Manual
powercarr freq.
offsetsinad
Query Example

Carrier power in dBm.
Carrier frequency in Hz.
Carrier frequency offset in Hz.
Signal to noise and distortion ratio in dB:CALC:DEM:AM:RES:MAX?
>9.840e+1,1.02e+2,-1.12e+1,3.21e+1,-1.14e+1,
2.7e+3,1.61e+1
S et
Query

:CALCulate:DEMod:AM:STATe
Description

Sets or queries the state of the AM Analysis function.

Syntax:CALCulate:DEMod:AM:STATe {ON|OFF|1|0}

Query Syntax:CALCulate:DEMod:AM:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurn AM Analysis off.
Turn AM Analysis on.
Turn AM Analysis off.
Turn AM Analysis on.
AM Analysis is off.
AM Analysis is on.

:CALC:DEM:AM:STAT 1S et:CALCulate:DEMod:EARPhone:STATe

Query

Description

Sets or queries the state of the ear phone out port.

Syntax:CALCulate:DEMod:EARPhone:STATe {ON|OFF|1|0}

Query Syntax:CALCulate:DEMod:EARPhone:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1

74Turn the phone output off.
Turn the phone output on.
Turn the phone output off.
Turn the phone output on.
Phone output is off.
Phone output is on.

エラー! スタイルが定義されていません。
Example:CALC:DEM:EARP:STAT 1:CALCulate:DEMod:FM:RESult:CURRent?

Query

Description

Returns the measurement results for FM
demodulation as a comma separated string.

Query Syntax:CALCulate:DEMod:FM:RESult:CURRent?

Return parameter <deviation,mode,rate,power,carr freq, offset,sinad>
Deviation
Frequency deviation in Hz.
mode
FM.
rate
Modulation rate in Hz.
power
Carrier power in dBm.
carr freq.
Carrier frequency in Hz.
offsetCarrier frequency offset in Hz.
sinad
Signal to noise and distortion ratio in dB
Query Example:CALC:DEM:FM:RES:CURR?
>9.840e+1,1.02e+2,-1.12e+1,3.21e+1,-1.14e+1,
2.7e+3,1.61e+1:CALCulate:DEMod: FM:RESult:MINimum?

Query

Description

Returns the minimum recorded measurement resultsfor FM demodulation as a comma separated string.

Query Syntax:CALCulate:DEMod:FM:RESult:MINimum?

Return parameter <deviation,mode,rate,power,carr freq, offset,sinad>
Deviation
Frequency deviation in %.
Mode
FM.
Rate
Modulation rate in Hz.
Power
Carrier power in dBm.
carr freq.
Carrier frequency in Hz.
OffsetCarrier frequency offset in Hz.
Sinad
Signal to noise and distortion ratio in dB

75

GSP-930 User Manual
Query Example:CALC:DEM:FM:RES:MIN?
>9.840e+1,1.02e+2,-1.12e+1,3.21e+1,-1.14e+1,
2.7e+3,1.61e+1:CALCulate:DEMod: FM:RESult:MAXimum?

Query

Description

Returns the maximum recorded measurement resultsfor FM demodulation as a comma separated string.

Query Syntax:CALCulate:DEMod:FM:RESult:MAXimum?

Return parameter <deviation,mode,rate,power,carr freq, offset,sinad>
Deviation
Frequency deviation in %.
mode
FM.
rate
Modulation rate in Hz.
power
Carrier power in dBm.
carr freq.
Carrier frequency in Hz.
offsetCarrier frequency offset in Hz.
sinad
Signal to noise and distortion ratio in dB
Query Example:CALC:DEM:FM:RES:MAX?
>9.840e+1,1.02e+2,-1.12e+1,3.21e+1,-1.14e+1,
2.7e+3,1.61e+1
S et:CALCulate:DEMod:FM:STATe

Query

Description

Sets or queries the state of the FM Analysis function.

Syntax:CALCulate:DEMod:FM:STATe {ON|OFF|1|0}

Query Syntax:CALCulate:DEMod:FM:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1
Example76

Turn FM Analysis off.
Turn FM Analysis on.
Turn FM Analysis off.
Turn FM Analysis on.
FM Analysis is off.
FM Analysis is on.

:CALC:DEM:FM:STAT 1

エラー! スタイルが定義されていません。
S et
Query

:CALCulate:JITTer:STATe
Description

Sets or queries the state of the Jitter Analysis
function.

Syntax:CALCulate:JITTer:STATe {ON|OFF|1|0}

Query Syntax:CALCulate:JITTer:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurn Jitter Analysis off.
Turn Jitter Analysis on.
Turn Jitter Analysis off.
Turn Jitter Analysis on.
Jitter Analysis is off.
Jitter Analysis is on.

:CALCulate:JITTer:STATe 1:CALCulate:JITTer:CARRier:POWer?
Description

Returns the carrier power in dBm.

Query Syntax:CALCulate:JITTer:CARRier:POWer?

Return parameter <NR3>
Query Example

Query

dBm

:CALC:JITT:CARR:POW?
>-5.237e+01:CALCulate:JITTer: PHASe?

Query

Description

Returns the carrier phase jitter in radians.

Query Syntax:CALCulate:JITTer:PHASe?

Return parameter <NR3>
Query Example

Rad:CALC:JITT:PHAS?
>1.5307e+01

77

GSP-930 User Manual

Query:CALCulate:JITTer:TIME?
Description

Returns the carrier jitter time in seconds.

Query Syntax:CALCulate:JITTer:TIME?

Return parameter <NR3>
Query Example

Seconds:CALC:JITT:TIME?
>.5.31e-08:CALCulate:LIMit<n>:CLEar

S et

Description

Clears the High Limit, Low Limit and the Pass/Failstate for the selected limit line.

Syntax:CALCulate:LIMit<n>:CLEar

Parameter<n>

Example:CALC:LIM1:CLE

Selected limit line

S et

:CALCulate:LIMit<n>:DATA

Query

Description

Sets or queries the frequency, amplitude limit ofevery point in the selected limit line. The data isstored in block data format.

Syntax:CALCulate:LIMit<n>:DATA <block data>

Query Syntax:CALCulate:LIMit<n>:DATA?

Parameter/<blockReturn parameter data><n>
ExampleSeleted limit line

:CALCulate:LIMit3:DATA?
>1e+6,-10,2e+6,-30,3e+6,-40,4e+6……………

:CALCulate:LIMit:FAIL?
78

Hz,dBm,Hz,dBm,Hz…………

Queryエラー! スタイルが定義されていません。
Description

Returns the Pass/Fail judgment.

Query Syntax:CALCulate:LIMit:FAIL?

Return parameter 0
1
Query Example

Pass
Fail

:CALC:LIM:FAIL?
>1S et:CALCulate:LIMit:LOW
Description

Selects which limit line is used for the low limit.

Syntax:CALCulate:LIMit:LOW <limit num>

Parameter<limit num> <NR1> 1~5

Example:CALC:LIM:LOW 2
S et:CALCulate:LIMit:HIGH
Description

Selects which limit line is used for the high limit.

Syntax:CALCulate:LIMit:HIGH <limit num>

Parameter<limit num> <NR1> 1~5

Example:CALC:LIM:HIGH 2:CALCulate:LIMit<n>:MARKer

S etDescription

Sets the current marker position to a point on a limit
line. The vertical position of the point is the marker’s
vertical position + a user-defined offset.

Syntax:CALCulate:LIMit<n>:MARKer <point>,<offset>

Parameter<point><offset><n>

Example:CALC:LIM1:MARK 5, 20<NR1> point 1~10
<NR3> dB
Seleted limit line79

GSP-930 User Manual
S et
Query

:CALCulate:LIMit:PASS:MODE
Description

Sets or queries the Pass/Fail conditions for the limit
line testing.

Syntax:CALCulate:LIMit:PASS:MODE {ALL|MAX|MIN}

Query Syntax:CALCulate:LIMit:PASS:MODE?

Parameter/
ALL
Return Parameter MAX
MIN
ExampleAll-in.
Max-InMin-In:CALC:LIM:PASS:MODE ALL
S et
Query

:CALCulate:LIMit:STATe
Description

Turns the current limit line on/off.

Syntax:CALCulate:LIMit:STATe {ON|OFF|1|0}

Query Syntax:CALCulate:LIMit:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleLimit line is off.
Limit line is on.
Limit line is off.
Limit line is on.
Limit line is off.
Limit line is on.

:CALC:LIM:STAT 1:CALCulate:LIMit<n>:TRACe

S etDescription

Creates a limit line from the current trace with a user
defined offset.

Syntax:CALCulate:LIMit<n>:TRACe <offset>80エラー! スタイルが定義されていません。
Parameter<n><offset><NR1> limit line 1~5
dB

Example:CALC:LIM2:TRAC 10:CALCulate:MARKer:AOFF
Description

Turns all the markers off.

Syntax:CALCulate:MARKer:AOFF

Example:CALC:MARK:AOFF

S et

S et

:CALCulate:MARKer<n>:FCOunt:RESolution

Query

Description

Sets or queries the frequency counter resolution in
Hz for the selected marker.

Syntax:CALCulate:MARKer<n>:FCOunt:RESolution <freq>

Query Syntax:CALCulate:MARKer<n>:FCOunt:RESolution?

Parameter<n><freq><NR1>Maker number 1~6<NR3> frequency resolution in Hz.

Return parameter <freq><NR3> frequency resolution in Hz.

Example:CALC:MARK1:FCO:RES?
>1.0e+3:CALCulate:MARKer<n>:FCOunt:
RESolution:AUTO

S et
Query

Description

Sets the frequency counter resolution Auto setting
on/off.

Syntax:CALCulate:MARKer<n>:FCOunt:RESolution:AUTO{ON|OFF|1|0}

Query Syntax:CALCulate:MARKer<n>:FCOunt:RESolution:AUTO?

81GSP-930 User Manual
Parameter

<n>
0
1
OFF
ON

Return parameter 0
1
ExampleMarker number 1~6
Auto is off.
Auto is on.
Auto is off.
Auto is on.
Auto is off.
Auto is on.

:CALC:MARK1:FCO:RES:AUTO?
S et:CALCulate:MARKer<n>:FCOunt:STATe

Query

Description

Sets or queries the state of the frequency counterfunction.

Syntax:CALCulate:MARKer<n>:FCOunt:STATe {ON|OFF|1|0}

Query Syntax:CALCulate:MARKer<n>:FCOunt:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurn frequency counter off.
Turn frequency counter on.
Turn frequency counter off.
Turn frequency counter on.
Frequency counter is off.
Frequency counter is on.

:CALC:MARKer1:FCO:STAT 1:CALCulate:MARKer<n>:FCOunt:X?

Query

Description

Returns the frequency of the selected marker in Hz.

Query Syntax:CALCulate:MARKer<n>:FCOunt:X?

Parameter<n>Return parameter <NR3>
Example82<NR1> Marker number 1~6
Frequency in Hz.

:CALC:MARK1:FCOU:X?
>2.0083e+8

エラー! スタイルが定義されていません。
S et
Query

:CALCulate:MARKer<n>:NOISe:STATe
Description

Sets or queries the state of the Marker Noisefunction.

Syntax:CALCulate:MARKer<n>:NOISe:STATe {ON|OFF|1|0}

Query Syntax:CALCulate:MARKer<n>:NOISe:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurn marker noise off.
Turn marker noise on.
Turn marker noise off.
Turn marker noise on.
Marker noise is off.
Marker noise is on.

:CALC:MARK2:NOIS:STAT ON
S et:CALCulate:MARKer<n>:NOISe:Y?

Query

Description

Returns the average noise level over a BW of 1Hz
from the marker position.

Query Syntax:CALCulate:MARKer<n>:NOISe:Y?

Parameter<n>Return parameter <NR3>
ExampleMaker number 1~6
Average noise level in dBm/Hz:CALC:MARK1:NOIS:Y?
>1.166e+2:CALCulate:MARKer<n>:PEAK

S etDescription

Sets the selected marker to the selected peak.

Query Syntax:CALCulate:MARKer<n>:PEAK {MAXimum|
MINimum|NEXT|RIGHt|LEFT}

Parameter<n>
MAXimum<NR1> Marker number 1~6
Highest peak value
83

GSP-930 User Manual
MIMimum
NEXT
RIGHt
LEFT
ExampleLowest peak value
Next peak
Next peak right
Next peak left

:CALC:MARK1:PEAK NEXT:CALCulate:MARKer<n>:PEAK:CTRack:
STATe

S et
Query

Description

Sets or queries the state of the Peak Track function.

Syntax:CALCulate:MARKer<n>:PEAK:CTRack:STATe{ON|OFF|1|0}

Query Syntax:CALCulate:MARKer<n>:PEAK:CTRack:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurn peak track off.
Turn peak track on.
Turn peak track off.
Turn peak track on.
Peak track is off.
Peak track is on.

:CALC:MARK1:PEAK:CTR:STAT ON:CALCulate:MARKer:PEAK:DATA?

Query

Description

Returns all of the peak data values??????????

Query syntax:CALCulate:MARKer:PEAK:DATA?

Return parameter <blockdata>
ExampleHz,dBm,Hz,dBm,Hz…………

:CALCulate:MARKer:PEAK:DATA?
>XXXXXXXXXXXXXx
S et:CALCulate:MARKer<n>:PEAK:EXCursion

84

Query

エラー! スタイルが定義されていません。
Description

Sets or queries the peak excursion value.

Syntax:CALCulate:MARKer<n>:PEAK:EXCursion <rel ampl>

Query Syntax:CALCulate:MARKer<n>:PEAK:EXCursion?

Parameter<n><rel ampl>

Return parameter <NR3>
ExampleMarker number 1~6
Peak excursion dB (offset fromthreshold)Peak excursion in dB.

:CALC:MARK1:PEAK:EXC 6 db
S et
Query

:CALCulate:MARKer:PEAK:SORT:TYPE
Description

Sets or queries the peak sort type for the peak table.

Syntax:CALCulate:MARKer:PEAK:SORT:TYPE{FREQuency|Amplitude}

Query Syntax:CALCulate:MARKer:PEAK:SORT:TYPE?

Parameter/
FREQuency Sort by frequency
Return parameter Amplitude Sort by amplitude
Example:CALC:MARK:PEAK:SORT:TYPE FREQ
S et

:CALCulate:MARKer:PEAK:TABLe:STATe

Query

Description

Sets or queries the state of the Peak Table.

Syntax:CALCulate:MARKer:PEAK:TABLe:STATe{ON|OFF|1|0}

Query Syntax:CALCulate:MARKer:PEAK:TABLe:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1Turn peak table off.
Turn peak table on.
Turn peak table off.
Turn peak table on.
peak table is off.
peak table is on.
85

GSP-930 User Manual
Example:CALC:MARK:PEAK:TABL:STAT ON
S et
Query

:CALCulate:MARKer:PEAK:THReshold
Description

Sets or queries the peak threshold value.

Syntax:CALCulate:MARKer:PEAK:THReshold < ampl>

Query Syntax
Parameter:CALCulate:MARKer:PEAK:THReshold?
< ampl>Peak Threshold level in mV/dBm

Return parameter <NR3>

ExamplePeak threshold. Note: the unit returned
depends on the currently set verticalunits.

:CALC:MARK:PEAK:THR -3 dBm:CALCulate:MARKer:PEAK:THReshold:
STATe

S et
Query

Description

Sets or queries the state of the Peak Threshold.

Syntax:CALCulate:MARKer:PEAK:THReshold:STATe{ON|OFF|1|0}

Query Syntax:CALCulate:MARKer:PEAK:THReshold:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurn peak threshold off.
Turn peak threshold on.
Turn peak threshold off.
Turn peak threshold on.
Peak threshold is off.
Peak threshold is on.

:CALC:MARK:PEAK:THR:STAT ON:CALCulate:MARKer<n>:SET
Description

86

S et

Sets the selected marker to one of five preset
positions.

エラー! スタイルが定義されていません。
Query Syntax:CALCulate:MARKer<n>:SET {CENTer|STARt|STOP|STEP|RLEVel}

Parameter<n>
CENTer
STARt
STOP
STEP
RLEVelExample:CALC:MARK1:SET CENT

<NR1> Marker number 1~6
Set to center frequency
Set to start frequency
Set to stop frequency
Set to CF STEP frequency
Set to the Reference level

S et:CALCulate:MARKer<n>:STATe

Query

Description

Sets or queries the state of the selected marker.

Syntax:CALCulate:MARKer<n>:STATe {ON|OFF|1|0}

Query Syntax:CALCulate:MARKer<n>:STATe?

Parameter

<n>
0
1
OFF
ON

Return parameter 0
1
Example<NR1> Marker number 1~6
Turn the selected marker off.
Turn the selected marker on.
Turn the selected marker off.
Turn the selected marker on.
The selected marker is off.
The selected marker on.

:CALC:MARK1:STAT ON
S et:CALCulate:MARKer:TABLe:STATe

Query

Description

Sets or queries the state of the marker table.

Syntax:CALCulate:MARKer:TABLe:STATe {ON|OFF|1|0}

Query Syntax:CALCulate:MARKer:TABLe:STATe

Parameter

01OFFTurn the table off.
Turn the table on.
Turn the table off.
87

GSP-930 User Manual
ON
Return parameter 0
1
ExampleTurn the table on.
The table is off.
The table is on.

:CALC:MARK:TABL:STAT ON
S et
Query

:CALCulate:MARKer<n>:TRACe
Description

Assigns a trace to the selected marker. Queries which
trace is assigned to the selected marker.

Syntax:CALCulate:MARKer<n>:TRACe <trace name>Query Syntax:CALCulate:MARKer<n>:TRACe?

Parameter/<n>Return parameter <tracename>Example<NR1> Marker number 1~6
The name of the trace: (A, B, C, D, E):CALC:MARK2:TRAC A
S et:CALCulate:MARKer<n>:TRACe:AUTO

Query

Description

Sets or queries the state of the Marker Tracefunction. Allows a trace to be automatically assigned
to the selected marker (on) or allows a trace to be
assigned manually (off).

Syntax:CALCulate:MARKer<n>:TRACe:AUTO {ON|OFF|1|0}

Query Syntax:CALCulate:MARKer<n>:TRACe:AUTO?

Parameter

<n>
0
1
OFF
ON

Return parameter 0
1
Example88<NR1> Marker number 1~6
Turn the auto function off.
Turn the auto function on.
Turn the auto function off.
Turn the auto function on.
The auto function is off.
The auto function is on.

:CALC:MARK2:TRAC:AUTO OFF

エラー! スタイルが定義されていません。
S et:CALCulate:MARKer<n>:TYPE

Query

Description

Sets or queries the marker type.

Syntax:CALCulate:MARKer<n>:TYPE {NORMal|DELTa}

Query Syntax:CALCulate:MARKer<n>:TYPE?

Parameter/<n><NR1> Marker number 1~6
Return parameter <NORMal> Normal marker<DELTa>Delta marker
Example:CALC:MARK1:TYPE NORM
S et
Query

:CALCulate:MARKer<n>:X
Description

Sets or returns the marker position in Hz.

Syntax:CALCulate:MARKer<n>:X <freq>

Query Syntax:CALCulate:MARKer<n>:X?

Parameter/<n>Return parameter <freq>
Example<NR1> Marker number 1~6
Hz:CALC:MARK4:X 2.0e+6
S et:CALCulate:MARKer<n>:Y?

Query

Description

Returns the marker vertical position in dBm.

Query Syntax:CALCulate:MARKer<n>:Y?

Parameter<n>Return parameter <NR3>
Example<NR1> Marker number 1~6
dBm:CALC:MARK1:Y?
>-5.43e+1

89

GSP-930 User Manual

S et:CALCulate:MATH:PDIF
Description

Calculates the power difference between two traces(TR1 and TR2).

Query Syntax:CALCulate:MATH:PDIF <Destination trace, TR1
trace,TR2 trace>

Parameter<des.
trace><TR1><TR2>

Example<NR1> Destination trace.
Source trace 1
Source trace 2:CALC:MATH:PDIF 1,2,3
S et:CALCulate:MATH:LDIF
Description

Calculates the logarithmic difference between two
traces (TR1 and TR2).

Query Syntax:CALCulate:MATH:LDIF <des. trace, TR1 trace,TR2
trace, Ref>

Parameter<des.
trace>
<TR1>
<TR2>
<Ref>

Example<NR1>Destination trace.
<NR1>Source trace 1<NR1>Source trace 2<NR1>Reference level:CALC:MATH:LDIF 1,2,3:CALCulate:MATH:LOFFS etDescription

Adds an offset to a source trace.

Query Syntax:CALCulate:MATH:LOFF <des. trace, TR1 trace,
offset>90

エラー! スタイルが定義されていません。
Parameter

Example<des.
trace><TR1><offset><NR1>Destination trace.
<NR1>Source trace 1<NR1>Offset in dB:CALC:MATH:LOFF 1,2,2.0e+1
S et
Query

:CALCulate:NDB:STATe
Description

Sets or queries the state of the NdB BW function.

Syntax:CALCulate:NDB:STATe {ON|OFF|1|0}

Query Syntax:CALCulate:NDB:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurn NdB BW off.
Turn NdB BW on.
Turn NdB BW off.
Turn NdB BW on.
NdB BW is off.
NdB BW is on.

:CALC:NDB:STAT ON
Query:CALCulate:NDB:BANDwidth|BWIDth?
Description

Returns the NdB bandwidth measurement.

Query Syntax:CALCulate:NDB:BANDwidth|BWIDth?

Return parameter <NR3>
ExampleNdB bandwidth in Hz.

:CALC:NDB:BAND?
>5.5e+04
S et:CALCulate:NORMalize:STATe

Query

Description

Turns the tracking generator normalization on/off.

Syntax:CALCulate:NORMalize:STATe{ON|OFF|1|0}91

GSP-930 User Manual
Query Syntax:CALCulate:NORMalize:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurn normalization off.
Turn normalization on.
Turn normalization off.
Turn normalization on.
normalization is off.
normalization is on.

:CALC:NORM:STAT ON
S et
Query

:CALCulate:OCBW:STATe
Description

Turns the OCBW measurement on/off.

Syntax:CALCulate:OCBW:STATe {ON|OFF|1|0}

Query Syntax:CALCulate:OCBW:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurn OCBW off.
Turn OCBW on.
Turn OCBW off.
Turn OCBW on.
OCBW is off.
OCBW is on.

:CALC:OCBW:STAT ON:CALCulate:OCBW:BANDwidth|BWIDth?

Query

Description

Returns the OCBW bandwidth measurement.

Query Syntax:CALCulate:OCBW:BANDwidth|BWIDth?

Return parameter <NR3>
ExampleOCBW bandwidth in Hz.

:CALC:OCBW:BAND?
>4.1e+03:CALCulate:OCBW:CHPower?
92

Query

エラー! スタイルが定義されていません。
Description

Returns the OCBW channel power measurement.

Query Syntax:CALCulate:OCBW:CHPower?

Return parameter <NR3>
ExampleOCBW channel power in dBm.

:CALC:OCBW:CHP?
>9.13e+01:CALCulate:OCBW:POWer?

Query

Description

Returns the OCBW total power measurement.

Query Syntax:CALCulate:OCBW:POWer?

Return parameter <NR3>
ExampleOCBW total power in dBm.

:CALC:OCBW:POW?
>1.33e+01:CALCulate:PMETer:POWer?

Query

Description

Returns the power meter power measurement.

Query Syntax:CALCulate:PMETer:POWer?

Return parameter <NR3>
ExamplePower in dBm.

:CALC:PMET:POW?
>-0.83e+01
S et:CALCulate:PMETer:LIMit:STATe

Query

Description

Turns the pass/fail limits on/off.

Syntax:CALCulate:PMETer:LIMit:STATe {ON|OFF|1|0}

Query Syntax:CALCulate:PMETer:LIMit:STATe?

Parameter

0
1
OFF
ON

Turns limits off.
Turns limits on.
Turns limits off.
Turns limits on.
93GSP-930 User Manual
Return parameter 0
1
ExampleTurns limits off.
Turns limits on.

:CALC:PMET:LIM:STAT?
>1
Query

:CALCulate:PMETer:LIMit:FAIL?
Description

Returns the pass/fail judgment.

Query Syntax:CALCulate:PMETer:LIMit:FAIL?

Return parameter 0
1
ExamplePass, or limits are not on.
Fail:CALC:PMET:LIM:FAIL?
>1
S et
Query

:CALCulate:SEM:STATe
Description

Turns the SEM measurement on/off.

Syntax:CALCulate:SEM:STATe {ON|OFF|1|0}

Query Syntax:CALCulate:SEM:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurns SEM off.
Turns SEM on.
Turns SEM off.
Turns SEM on.
SEM is off.
SEM is on.

:CALC:SEM:STAT ON:CALCulate:SEM:OFFSet<n>:RESult?

Query

Description

Returns the start, stop frequencies as well as thepass/fail limits and judgements for the chosen offset.

Query syntax:CALCulate:SEM:OFFSet<n>:RESult?

94エラー! スタイルが定義されていません。
Parameter<n>Offset number 1~5.

Return parameter <start freq> Start frequency of the selected channel<low dBm> Lower dBm measurement<low p/f> Lower pass/fail limit judgment.
0 = pass, 1 = fail.
<stop freq> Stop frequency of the selected channel<upp dBm> Upper dBm measurement<upp p/f> Upper pass/fail limit judgment
0 = pass, 1 = fail.
Example:CALC:SEM:OFFS<n>:RES?
>9e+7, -7.9e+1,0,1.7e+7,-6.9e+1,0
Query:CALCulate:TOI:DIFFerential?
Description

Returns the third order intermodulation distortion.

Query syntax:CALCulate:TOI:DIFFerential?

Return parameter <base lower>dBc<base upper>
dBc<3rd order lower> dBc
Example:CALC:TOI:DIFF?
>0.0e+0,-1.67e-1,-1.09e+1,-6.61e+0
Query:CALCulate:TOI:FREQuency:STEPsize?
Description

Returns the delta of the base lower and base upper.

Query syntax:CALCulate:TOI:FREQuency:STEPsize?

Return parameter <NR3>
ExampleΔf: Hz:CALC:TOI:FREQ:STEP?
>6.65e+5
S et:CALCulate:TOI:LIMit:STATe

Query

95

GSP-930 User Manual
Description

Turns the TOI pass/fail limit on/off.

Syntax:CALCulate:TOI:LIMit:STATe {ON|OFF|1|0}

Query Syntax:CALCulate:TOI:LIMit:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurns pass/fail limit off.
Turns pass/fail limit on.
Turns pass/fail limit off.
Turns pass/fail limit on.
Turns pass/fail limit off.
Turns pass/fail limit on.

:CALC:TOI:LIM:STAT ON
Query:CALCulate:TOI:RESult?
Description

Returns the third order intercept and the pass/failjudgments.

Query syntax:CALCulate:TOI:RESult?

Return parameter <3rd lower><lower p/f>
<3rd upper><upper p/f>
Example3rd order lower intercept
3rd order lower pass/fail judgment.
0=pass, 1=fail3rd order upper intercept
3rd order lower pass/fail judgment
0=pass, 1=fail:CALC:TOI:RES?
>-5.5e+1,0, -6.61e+1,0
S et
Query

:CALCulate:TOI:STATe
Description

Turns TOI measurement on/off.

Syntax:CALCulate:TOI:STATe {ON|OFF|1|0}

Query Syntax:CALCulate:TOI:STATe?

Parameter

0196

Turns TOI measurement off.
Turns TOI measurement on.

エラー! スタイルが定義されていません。
OFFON
Return parameter 0
1
ExampleTurns TOI measurement off.
Turns TOI measurement on.
TOI measurement is off.
TOI measurement is on.

:CALC:TOI:STAT ON

CONFigure Commands:CONFigure:MODE ................................................................... 97
S et
Query

:CONFigure:MODE
Description

Sets or queries the operating mode, spectrum or
power meter.

Query Syntax:CONFigure:MODE {SA|PMET}

Parameter/<SA>
Return parameter <PMET>
ExampleSpectrum mode
Power meter mode

:CONF:MODE SA

DISPlay Commands:DISPlay:BRIGhtness............................................................... 98:DISPlay:ENABle ....................................................................... 98:DISPlay:DEMod[:WINDow]:TRACe:X[:SCALe]:AUTO 99:DISPlay:DEMod[:WINDow]:TRACe:X[:SCALe]:PDIVision .................................................................................... 99:DISPlay:DEMod[:WINDow]:TRACe:X[:SCALe]:RPOSition..............................................................100:DISPlay:DEMod[:WINDow]:TRACe:X[:SCALe]:RVALue10:DISPlay:DEMod[:WINDow]:TRACe:Y[:SCALe]:AUTO101:DISPlay:DEMod[:WINDow]:TRACe:Y[:SCALe]:PDIVision ...............................................................101:DISPlay:DEMod[:WINDow]:TRACe:Y
97

GSP-930 User Manual[:SCALe]:RPOSition.............................................................. 101:DISPlay:DEMod[:WINDow]:TRACe:Y[:SCALe]:RVALue10:DISPlay[:WINDow]:NORMal............................................... 102:DISPlay[:WINDow]:SPECtrogram.................................... 102:DISPlay[:WINDow]:SPLit:NORMal .................................. 103:DISPlay[:WINDow]:SPLit:NORMal:ALTernate............ 103:DISPlay[:WINDow]:SPLit:NORMal:TYPE...................... 103:DISPlay[:WINDow]:SPLit:SPECtrogram ....................... 103:DISPlay[:WINDow]:SPLit:TOPO ...................................... 104:DISPlay[:WINDow]:SPLit:TOPO ...................................... 104:DISPlay[:WINDow]:TRACe<n>:MODE ........................... 104:DISPlay[:WINDow]:TRACe:Y:DLINe ............................... 104:DISPlay[:WINDow]:TRACe:Y:DLINe:STATe................ 105:DISPlay[:WINDow]:TRACe:Y[:SCALe]:NRLevel ........ 105:DISPlay[:WINDow]:TRACe:Y[:SCALe]:NRPosition .. 106:DISPlay[:WINDow]:TRACe:Y[:SCALe]:PDIVision ..... 106:DISPlay[:WINDow]:TRACe:Y[:SCALe]:POSition....... 106:DISPlay[:WINDow]:TRACe:Y[:SCALe]:RLEVel .......... 107:DISPlay[:WINDow]:TRACe:Y[:SCALe]:RLEVel:OFFSet10:DISPlay[:WINDow]:TRACe:Y[:SCALe]:SPACing ....... 108:DISPlay[:WINDow]:TRACe:Y[:SCALe]:STATe........... 108
S et
Query

:DISPlay:BRIGhtness
Description

Sets or queries the LCD brightness level.

Syntax:DISPlay:BRIGhtness {HIGH|MIDDle|LOW}

Query Syntax:DISPlay:BRIGhtness?

Parameter/<HIGH>
Return parameter <MIDDle><LOW>
ExampleHigh brightness level
Mid brightness level
Low brightness level:DISP:BRIG HIGHS et

:DISPlay:ENABle

98

Query

エラー! スタイルが定義されていません。
Description

Turns the LCD backlight on/off.

Syntax:DISPlay:ENABle {OFF|ON|0|1}

Query Syntax:DISPlay:ENABle?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurn LCD backlight off.
Turn LCD backlight on.
Turn LCD backlight off.
Turn LCD backlight on.
LCD backlight is off.
LCD backlight is on.

DISP:ENAB?
>1:DISPlay:DEMod[:WINDow]:TRACe:X[:SCALe]:AUTO

S et
Query

Description

Turns auto scale on/off for the x axis for AM/FM
demodulation.

Syntax:DISPlay:DEMod[:WINDow]:TRACe:X[:SCALe]:AUTO{OFF|ON|0|1}

Query Syntax:DISPlay:DEMod[:WINDow]:TRACe:X[:SCALe]:AUTO?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurn Auto Scale off.
Turn Auto Scale on.
Turn Auto Scale off.
Turn Auto Scale on.
Auto Scale is off.
Auto Scale is on.

:DISP:DEM:TRAC:X:AUTO ON:DISPlay:DEMod[:WINDow]:TRACe:X[:SCALe]:PDIVision
Description

S et
Query

Sets or queries the Time axis Scale/Div.
99

GSP-930 User Manual
Syntax:DISPlay:DEMod[:WINDow]:TRACe:X[:SCALe]:
PDIVision <time>

Query Syntax:DISPlay:DEMod[:WINDow]:TRACe:X[:SCALe]:
PDIVision?

Parameter/<time>
Return parameter
Example<NR3>Time division. Default = seconds:DISP:DEM:TRAC:X:PDIV 2 ms

:DISPlay:DEMod[:WINDow]:TRACe:X[:SCALe]:RPOSition

S et
Query

Description

Sets or queries the Reference Position of the trace
for AM/FM demodulation (x-axis grid division).

Syntax:DISPlay:DEMod[:WINDow]:TRACe:X[:SCALe]:
RPOSition <integer>

Query Syntax:DISPlay:DEMod[:WINDow]:TRACe:X[:SCALe]:
RPOSition?

Parameter/<integer>
Return parameter
Example<NR1>1~10:DISP:DEM:TRAC:X:RPOS 2:DISPlay:DEMod[:WINDow]:TRACe:X[:SCALe]:RVALue

S et
Query

Description

Sets or queries the Reference value time.

Syntax:DISPlay:DEMod[:WINDow]:TRACe:X[:SCALe]:
RVALue <time>

Query Syntax:DISPlay:DEMod[:WINDow]:TRACe:X[:SCALe]:
RVALue?

100

エラー! スタイルが定義されていません。
Parameter/<time>
Return parameter
Example<NRf>:DISP:DEM:TRAC:X:RVAL 2 ms

:DISPlay:DEMod[:WINDow]:TRACe:Y[:
SCALe]:AUTO

S etDescription

Sets the vertical display scale to auto for AM/FM
demodulation.

Syntax:DISPlay:DEMod[:WINDow]:TRACe:Y[:SCALe]:
AUTO <integer>

Example:DISP:DEM:TRAC:Y:AUTO:DISPlay:DEMod[:WINDow]:TRACe:Y[:SCALe]:PDIVision

S et
Query

Description

Sets or queries the Y-axis scale division.

Syntax:DISPlay:DEMod[:WINDow]:TRACe:Y[:SCALe]:
PDIVision <integer>

Query Syntax:DISPlay:DEMod[:WINDow]:TRACe:Y[:SCALe]:
PDIVision?

Parameter/<integer>
Return parameter
Example<NR1> AM Unit: %, FM Unit: Hz:DISP:DEM:TRAC:Y:PDIV 20:DISPlay:DEMod[:WINDow]:TRACe:Y[:SCALe]:RPOSition
Description

S et
Query

Sets or queries the Reference Position of the trace
for AM/FM demodulation (y-axis grid division).

101

GSP-930 User Manual
Syntax:DISPlay:DEMod[:WINDow]:TRACe:Y[:SCALe]:
RPOSition <integer>

Query Syntax:DISPlay:DEMod[:WINDow]:TRACe:Y[:SCALe]:
RPOSition?

Parameter/<integer>
Return parameter
Example<NR1>1~10:DISP:DEM:TRAC:Y:RPOS 2:DISPlay:DEMod[:WINDow]:TRACe:Y[:SCALe]:RVALue

S et
Query

Description

Sets or queries the Reference value(AM: %, FM: Hz).

Syntax:DISPlay:DEMod[:WINDow]:TRACe:Y[:SCALe]:
RVALue <integer>

Query Syntax:DISPlay:DEMod[:WINDow]:TRACe:Y[:SCALe]:
RVALue?

Parameter/<integer>
Return parameter
Example<NR1> FM Unit: Hz, AM Unit: %:DISP:DEM:TRAC:Y:RVAL 2 %:DISPlay[:WINDow]:NORMal

S et

Description

Sets the display window to spectrum mode.

Syntax:DISPlay[:WINDow]:NORMal

Example:DISP:NORM:DISPlay[:WINDow]:SPECtrogram
Description

102

S et

Sets the display window to spectrogram mode.

エラー! スタイルが定義されていません。
Syntax:DISPlay[:WINDow]:SPECtrogram

Example:DISP:SPEC:DISPlay[:WINDow]:SPLit:NORMal

S et

Description

Creates two split windows, both in spectrum mode.

Syntax:DISPlay[:WINDow]:SPLit:NORMal

Example:DISP:SPL:NORM:DISPlay[:WINDow]:SPLit:NORMal:
ALTernate

S et

Description

Turns on Alternate Sweep function for split window
mode.

Syntax:DISPlay[:WINDow]:SPLit:NORMal:ALTernate

Example:DISP:SPL:NORM:ALT:DISPlay[:WINDow]:SPLit:NORMal:TYPE

S et

Description

Sets the active window in split screen mode.

Syntax:DISPlay[:WINDow]:SPLit:NORMal:TYPE{UPPer|LOWer}

Example:DISP:SPL:NORM:TYPE UPP:DISPlay[:WINDow]:SPLit:SPECtrogram

S etDescription

Sets the split screen mode to Spectrogram +Spectrum.

Syntax:DISPlay[:WINDow]:SPLit:SPECtrogram

Example:DISP:SPL:SPEC

103

GSP-930 User Manual:DISPlay[:WINDow]:SPLit:TOPO

S etDescription

Sets the split screen mode to Topographic +Spectrum.

Syntax:DISPlay[:WINDow]:SPLit:TOPO

Example:DISP:SPL:TOPOS et:DISPlay[:WINDow]:TOPO
Description

Sets the display window to topographic.

Syntax:DISPlay[:WINDow]:TOPO

Example:DISP:TOPO:DISPlay[:WINDow]:TRACe<n>:MODE

S etDescription

Sets or queries the marker type.

Syntax:DISPlay[:WINDow]:TRACe<n>:MODE{WRITe|VIEW|BLANk|MAXHold|MINHold|PHOLd}

Parameter

<n>
WRITe
VIEW
BLANK
MAXHold
MINHold
PHOLd

Example:DISP:TRAC4:MODE VIEW

<NR1> Trace number 1~4
Clear and Write
Hold the last trace
Clears the trace
Hold the maximum/minimum points fromeach trace
WHAT is PHOLD?

S et:DISPlay[:WINDow]:TRACe:Y:DLINe

Query

Description

Sets the display line amplitude level.

Syntax:DISPlay[:WINDow]:TRACe:Y:DLINe <ampl>

104エラー! スタイルが定義されていません。
Query Syntax:DISPlay[:WINDow]:TRACe:Y:DLINe?

Parameter/<ampl>
Return parameter
Example<NR3> dBm

:DISP:TRAC:Y:DLIN -5.0e+1
S et:DISPlay[:WINDow]:TRACe:Y:DLINe:STATe

Query

Description

Turns the display line on/off.

Syntax:DISPlay[:WINDow]:TRACe:Y:DLINe:STATe{OFF|ON|0|1}

Query Syntax:DISPlay[:WINDow]:TRACe:Y:DLINe:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurn display line off.
Turn display line on.
Turn display line off.
Turn display line on.
The display line is off.
The display line is on.

:DISP:TRAC:Y:DLIN:STAT ON:DISPlay[:WINDow]:TRACe:Y[:SCALe]:
NRLevel

S et
Query

Description

Sets or queries the normalized reference level for the
TG option.

Syntax:DISPlay[:WINDow]:TRACe:Y[:SCALe]:NRLevel<rel_ampl>

Query Syntax:DISPlay[:WINDow]:TRACe:Y[:SCALe]:NRLevel?

Parameter/<rel_ampl>
Return parameter
Example<NR3> dB

:DISP:TRAC:Y:NRLevel 5 dB

105

GSP-930 User Manual:DISPlay[:WINDow]:TRACe:Y[:SCALe]:
NRPosition

S et
Query

Description

Sets the position of the normalized reference level.
The 0~10 Y-axis grid divisions correspond to thebottom~top grid divisions.

Syntax:DISPlay[:WINDow]:TRACe:Y[:SCALe]:NRPosition<integer>

Query Syntax:DISPlay[:WINDow]:TRACe:Y[:SCALe]:NRPosition?

Parameter/<integer>
Return parameter
Example<NR1> 0~10

:DISP:TRAC:Y:NRP 5:DISPlay[:WINDow]:TRACe:Y[:SCALe]:PDIVi
sion

S et
Query

Description

Sets or queries the Y-axis scale/div when theamplitude scale is logarithmic.

Syntax:DISPlay[:WINDow]:TRACe:Y[:SCALe]:PDIVision{1|2|5|
10}

Query Syntax:DISPlay[:WINDow]:TRACe:Y[:SCALe]:PDIVision?

Parameter/
1
Return parameter 2
5
10
Example

1 dBm
2 dBm
5 dBm
10 dBm:DISP:TRAC:Y:PDIV 10:DISPlay[:WINDow]:TRACe:Y[:SCALe]:
POSition
Description
106

S et
Query

Sets or queries the position of the on-screen scale.

エラー! スタイルが定義されていません。
Syntax:DISPlay[:WINDow]:TRACe:Y[:SCALe]:POSition{LEFT|CENTer|RIGHt}

Query Syntax:DISPlay[:WINDow]:TRACe:Y[:SCALe]:POSition?

Parameter/
LEFT
Return parameter CENTer
RIGHt
ExamplePosition the scale to left
Position the scale to the center
Position the scale to right:DISP:TRAC:Y:POS LEFT:DISPlay[:WINDow]:TRACe:Y[:SCALe]:
RLEVelS etQuery

Description

Sets or queries the Y-axis reference level. The units
depend on the scale type (logarithmic/linear).

Syntax:DISPlay[:WINDow]:TRACe:Y[:SCALe]:RLEVel <ampl>

Query Syntax:DISPlay[:WINDow]:TRACe:Y[:SCALe]:RLEVel?

Parameter/<ampl>
Return parameter
Example<NR3>:DISP:TRAC:Y:RLEV 1 mV:DISPlay[:WINDow]:TRACe:Y[:SCALe]:
RLEVel:OFFSet

S et
Query

Description

Sets or queries the Y-axis reference level offset.

Syntax:DISPlay[:WINDow]:TRACe:Y[:SCALe]:RLEVel:OFFSet<rel_ampl>

Query Syntax:DISPlay[:WINDow]:TRACe:Y[:SCALe]:RLEVel:OFFSet?

Parameter/<ampl>
Return parameter
Example<NR3> dB

:DISP:TRAC:Y:RLEV OFFS -5.0e+1 dB

107

GSP-930 User Manual:DISPlay[:WINDow]:TRACe:Y[:SCALe]:SPACing

S et
Query

Description

Sets or queries the type of scale: logarithmic orlinear.

Syntax:DISPlay[:WINDow]:TRACe:Y[:SCALe]:SPACing{LINear|LOGarithmic}

Query Syntax:DISPlay[:WINDow]:TRACe:Y[:SCALe]:SPACing?

Parameter/
LINearReturn parameter LOGarithmic
ExampleLinear scale
Logarithmic scale:DISP:TRAC:Y:SPAC LOG:DISPlay[:WINDow]:TRACe:Y[:SCALe]:
STATe

S et
Query

Description

Turns the on-screen scale on/off.

Syntax:DISPlay[:WINDow]:TRACe:Y[:SCALe]:STATe{OFF|ON|0|1}

Query Syntax:DISPlay[:WINDow]:TRACe:Y[:SCALe]:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurn scale off.
Turn scale on.
Turn scale off.
Turn scale on.
Scale is off.
Scale is on.

:DISP:TRAC:Y:STAT ON

FORMat Commands:FORMat:TRACe:DATA........................................................ 109

108

エラー! スタイルが定義されていません。
S et
Query

:FORMat:TRACe:DATA
Description

Sets or queries data format used to save trace data.

Syntax:FORMat:TRACe:DATA{ASC|BINary|HEXadecimal|INTeger|OCTal|REAL}

Query Syntax:FORMat:TRACe:DATA?

Parameter/
ASC
Return parameter BINary
HEXadecimal
INTeger
OCTal
REAL
ExampleASCII???
BinaryHexadecimal
Integer
Octal
Real number data

:FORM:TRAC:DATA ASC

109

GSP-930 User Manual

INITiate Commands:INITiate:CONTinuous........................................................... 110:INITiate[:IMMediate] ............................................................ 110
S et
Query

:INITiate:CONTinuous
Description

Set the trigger to continuous or single.
ON=continuous, OFF=single trigger. When set to OFF
the trigger is initiated with the “:INITiate[:IMMediate]”
command.

Syntax:INITiate:CONTinuous {OFF|ON|0|1}

Query Syntax:INITiate:CONTinuous?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleSet trigger to single.
Set trigger to continuous.
Set trigger to single.
Set trigger to continuous.
Set trigger to single.
Set trigger to continuous.

:INIT:CONT ON:INITiate[:IMMediate]

S et

Description

Initiates an immediate trigger when the trigger is set
to single (:INITiate:CONTinuous OFF).

Syntax:INITiate[:IMMediate]

Example

:INIT

110

エラー! スタイルが定義されていません。

INPut Commands:INPut:ATTenuation ...............................................................111:INPut:ATTenuation:AUTO..................................................111:INPut:IMPedance...................................................................112:INPut:OFFSet .........................................................................112
S et
Query

:INPut:ATTenuation
Description

Sets or queries the input attenuation.

Syntax:INPut:ATTenuation <rel_ampl>

Query Syntax:INPut:ATTenuation?

Parameter/<rel_ampl>
Return parameter
Example<NR3> dB

:INP:ATT 10 dB
S et
Query

:INPut:ATTenuation:AUTO
Description

Sets or queries whether the automatic input
attenuation is on/off.

Syntax:INPut:ATTenuation:AUTO {OFF|ON|0|1}

Query Syntax:INPut:ATTenuation:AUTO?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurn automatic input attenuation off.
Turn automatic input attenuation on.
Turn automatic input attenuation off.
Turn automatic input attenuation on.
Automatic input attenuation is off.
Automatic input attenuation is on.

:INP:ATT ON

111

GSP-930 User Manual
S et
Query

:INPut:IMPedance
Description

Sets or the input impedance in Ω.

Syntax:INPut:IMPedance {50|75}

Query Syntax:INPut:IMPedance?

Parameter/
50
Return parameter 75
Example

<NR1>
<NR1>

:INP:IMP 75
S et
Query

:INPut:OFFSet
Description

Sets or queries the input offset (Input Z Calibration).

Syntax:INPut:OFFSet <rel_ampl>

Query Syntax:INPut:OFFSet?

Parameter/<rel_ampl>
Return parameter
Example112<NR3> dB

:INP:OFFS 10 dB

エラー! スタイルが定義されていません。

MMEMory Commands:MMEMory:CATalog? .............................................................113:MMEMory:CDIRectory.........................................................113:MMEMory:COPY ....................................................................114:MMEMory:DELete .................................................................114:MMEMory:DESTination .......................................................114:MMEMory:LOAD:CORRection..........................................115:MMEMory:LOAD:LIMit .........................................................115:MMEMory:LOAD:PMET.......................................................115:MMEMory:LOAD:SEQuence..............................................115:MMEMory:LOAD:STATe .....................................................116:MMEMory:LOAD:TRACe.....................................................116:MMEMory:MOVE....................................................................116:MMEMory:REName ...............................................................117:MMEMory:STORe:CORRection .......................................117:MMEMory:STORe:LIMit.......................................................117:MMEMory:STORe:PMET.....................................................117:MMEMory:STORe:SCReen ................................................118:MMEMory:STORe:SEQuence ...........................................118:MMEMory:STORe:STATe...................................................118:MMEMory:STORe:TRACe ..................................................119:MMEMory:CATalog?

Query

Description

Returns a list of all the files that have been saved to
the local memory.

Query Syntax:MMEMory:CATalog?

Example:MMEM:CAT?
>LocalState1.sta, QuickJpg.jpg,QuickJpg1.jpg,………

:MMEMory:CDIRectory
Description

S et

Sets the source directory.
113

GSP-930 User Manual
Syntax:MMEMory:CDIRectory {LOCAL|USB|SD}

Parameter

LOCAL
USB
SD

Example:MMEM:CD LOCAL:MMEMory:COPY

S et

Description

Copies a designated file from the current file
directory to the destination directory. The file can berenamed after it is copied.

Syntax:MMEMory:COPY <src_file_name>,<dest_file_name>Parameter<src_file_name><dest_file_name>Example:MMEM:COPY quick1.jpg, quick2.jpg:MMEMory:DELete

S et

Description

Deletes the designated file from the current directory.

Syntax:MMEMory:DELete <src_file_name>Parameter<src_file_name>Example:MMEM:DEL quick1.jpg:MMEMory:DESTination

S et

Description

Sets the source directory.

Syntax:MMEMory:DESTination {LOCAL|USB|SD}

Parameter

LOCAL
USB
SD

Example:MMEM:DEST SD

114

エラー! スタイルが定義されていません。

S et:MMEMory:LOAD:CORRection
Description

Loads correction data from a file to the internal
memory.

Syntax:MMEMory:LOAD:CORRection <corr num>,<dest_file_name>Parameter<corr num><dest_file_name><NR1> correction set 1~5
XXX.cor

Example:MMEM:LOAD:CORR 2,test.corr
S et:MMEMory:LOAD:LIMit
Description

Loads limit line data from a file to the internal
memory.

Syntax:MMEMory:LOAD:LIMit <lim num>,<dest_file_name>Parameter<lim num><dest_file_name>Example:MMEM:LOAD:LIM 2,test.lim<NR1> limit line 1~5
XXX.lim

S et:MMEMory:LOAD:PMET
Description

Loads power meter data from a file to the internal
memory.

Syntax:MMEMory:LOAD:PMET <dest_file_name>Parameter<dest_file_name>Example:MMEMory:LOAD:PMET test.pmet:MMEMory:LOAD:SEQuence
Description

XXX.pmet

S etLoads sequence data from a file to the internal
memory.
115

GSP-930 User Manual
Syntax:MMEMory:LOAD:SEQuence <seq num>,<dest_file_name>Parameter<seq num><dest_file_name>Example:MMEM:LOAD:SEQ 2,test.seq<NR1>sequence number 1~5
XXX.seq

S et:MMEMory:LOAD:STATe
Description

Loads the instrument state from a file to the internal
memory.

Syntax:MMEMory:LOAD:STATe <dest_file_name>Parameter<dest_file_name>Example:MMEM:LOAD:STAT test.stat

XXX.stat

S et:MMEMory:LOAD:TRACe
Description

Loads trace data from a file to the internal memory.

Syntax:MMEMory:LOAD:TRACe <trace name>,<dest_file_name>Parameter<trace name><dest_file_name>Example:MMEM:LOAD:TRAC 2,test.tra:MMEMory:MOVE<NR1> 1~4
XXX.traS et

Description

Moves a designated file from the current file directory
to the destination directory. The file can be renamed
after it is moved.

Syntax:MMEMory:MOVE <src_file_name>,<dest_file_name>Parameter<src_file_name><dest_file_name>116エラー! スタイルが定義されていません。
Example:MMEM:MOVE quick1.jpg, quick2.jpg
S et:MMEMory:REName
Description

Renames the designated file from the current file.

Syntax:MMEMory:REName <old_file_name>,<new_file_name>Parameter<old_file_name><new_file_name>Example:MMEM:REN quick1.jpg, quick2.jpg
S et:MMEMory:STORe:CORRection
Description

Store correction data to a file from the internal
memory.

Syntax:MMEMory:STOR:CORRection<corr num>,<new_dest_file_name>Parameter<corr num>
<new_dest_file_name>Example:MMEM:STOR:CORR 2,test.corr<NR1> correction set 1~5
XXX.cor

S et:MMEMory:STORe:LIMit
Description

Store limit line data to a file from the internal
memory.

Syntax:MMEMory:STOR:LIMit<lim num>,<new_dest_file_name>Parameter<lim num><new_dest_file_name>Example:MMEM:STOR:LIM 2,test.lim:MMEMory:STORe:PMET<NR1> limit line 1~5
XXX.limS et

117

GSP-930 User Manual
Description

Store power meter data to a file from the internal
memory.

Syntax:MMEMory:STORe:PMET <dest_file_name>Parameter<dest_file_name>Example:MMEMory:STOR:PMET test.pmet

XXX.pmet

S et:MMEMory:STORe:SCReen
Description

Store a screen-shot to the current file directory.

Syntax:MMEMory:STORe:SCReen <new_dest_file_name>Parameter<new_dest_file_name>Example:MMEMory:STOR:SCR test.jpg

XXX.jpg

S et:MMEMory:STORe:SEQuence
Description

Store sequence data to a file from the internal
memory.

Syntax:MMEMory:STORe:SEQuence <seq num>,<new_dest_file_name>Parameter<seq num><new_dest_file_name>Example<NR1>sequence number
1~5
XXX.seq:MMEM:STOR:SEQ 2,test.seq
S et:MMEMory:STORe:STATe
Description

Store the instrument state to a file from the internal
memory.

Syntax:MMEMory:STORe:STATe <new_dest_file_name>Parameter<new_dest_file_name>Example:MMEM:STORe:STAT test.stat

118

XXX.statエラー! スタイルが定義されていません。

S et:MMEMory:STORe:TRACe
Description

Store trace data to a file from the internal memory.

Syntax:MMEMory:STORe:TRACe <trace name>,<new_dest_file_name>Parameter<trace name><new_dest_file_name>Example:MMEM:STORe:TRAC 2,test.tra<NR1> 1~4
XXX.tra

OUTPut Commands:OUTPut[:STATe]...................................................................119
S et
Query

:OUTPut[:STATe]
Description

Turns the tracking generator output on/off.

Syntax:OUTPut[:STATe] {OFF|ON|0|1}

Query Syntax:OUTPut[:STATe]?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurn TG output off.
Turn TG output on.
Turn TG output off.
Turn TG output on.
TG output is off.
TG output is on.

:OUTP ON119

GSP-930 User Manual

SENSe Commands[:SENSe]:ACPR:ACHannel<n>:BANDwidth|BWIDth . 123[:SENSe]:ACPR:ACHannel<n>:HLIMit............................ 123[:SENSe]:ACPR:ACHannel<n>:LLIMit ............................ 124[:SENSe]:ACPR:ACHannel<n>:OFFSet ......................... 124[:SENSe]:ACPR:BANDwidth|BWIDth.............................. 124[:SENSe]:ACPR:HLIMit ........................................................ 125[:SENSe]:ACPR:LLIMit......................................................... 125[:SENSe]:ACPR:HELP .......................................................... 125[:SENSe]:ACPR:SPACe....................................................... 126[:SENSe]:ASET:AMPLitude................................................ 126[:SENSe]:ASET:AMPLitude:AUTO .................................. 126[:SENSe]:ASET:RUN ............................................................. 127[:SENSe]:ASET:SPAN .......................................................... 127[:SENSe]:ASET:SPAN:AUTO............................................. 127[:SENSe]:AVERage:COUNt ................................................ 128[:SENSe]:AVERage:STATe................................................. 128[:SENSe]:AVERage:TYPE ................................................... 128[:SENSe]:BANDwidth|BWIDth[:RESolution]................. 129[:SENSe]:BANDwidth|BWIDth[:RESolution]:AUTO... 129[:SENSe]:BANDwidth|BWIDth:VIDeo.............................. 129[:SENSe]:BANDwidth|BWIDth:VIDeo:AUTO ................ 130[:SENSe]:CHANnel:SPACe:DOWN .................................. 130[:SENSe]:CHANnel:SPACe:UP.......................................... 130[:SENSe]:CNR:CHANnel:SPACe ...................................... 131[:SENSe]:CNR:DELTamarker:MODE............................... 131[:SENSe]:CORRection:CSET<n>:DATA ........................ 131[:SENSe]:CORRection:CSET<n>:STATe ...................... 132[:SENSe]:CORRection:CSET<n>:DELete ..................... 132[:SENSe]:CSO:CHANnel:SPACe ...................................... 132[:SENSe]:CTB:CHANnel:SPACe ...................................... 133[:SENSe]:DEMod:EARPone:TYPE.................................... 133[:SENSe]:DEMod:EARPone:VOLume.............................. 133[:SENSe]:DEMod:EARPone:GAIN .................................... 134[:SENSe]:DEMod:FILTer:LPASs ....................................... 134[:SENSe]:DEMod:IFBW......................................................... 135
120

エラー! スタイルが定義されていません。
[:SENSe]:DETector[:FUNCtion]........................................135[:SENSe]:DETector[:FUNCtion]:AUTO..........................135[:SENSe]:EMIFilter:STATe..................................................136[:SENSe]:EMIFilter:BANDwidth|BWIDth[:RESolution]136[:SENSe]:FREQuency:CENter ...........................................137[:SENSe]:FREQuency:CENter:STEP...............................137[:SENSe]:FREQuency:CENter:STEP:AUTO.................137[:SENSe]:FREQuency:OFFSet ..........................................138[:SENSe]:FREQuency:SPAN..............................................138[:SENSe]:FREQuency:SPAN:FULL..................................138[:SENSe]:FREQuency:SPAN:PREVious.........................138[:SENSe]:FREQuency:STARt ............................................139[:SENSe]:FREQuency:STOP..............................................139[:SENSe]:LIMit<n>:DELete..................................................139[:SENSe]:JITTer:OFFSet:STARt......................................140[:SENSe]:JITTer:OFFSet:STOP .......................................140[:SENSe]:NDB:BANDwidth|BWIDth .................................140[:SENSe]:OCBW:BANDwidth|BWIDth .............................140[:SENSe]:OCBW:PERCent ..................................................141[:SENSe]:OCBW:SPACe ......................................................141[:SENSe]:OCBW:SPAN.........................................................141[:SENSe]:PMETer:FREQuency..........................................142[:SENSe]:PMETer:HLIMit.....................................................142[:SENSe]:PMETer:HOLD:STATe ......................................142[:SENSe]:PMETer:LLIMit.....................................................143[:SENSe]:PMETer:PSENsor:MODE .................................143[:SENSe]:PMETer:RECording:TIME.................................143[:SENSe]:PMETer:RECording:TIME:STEP ....................144[:SENSe]:POWer[:RF]:GAIN ...............................................144[:SENSe]:SEMask:BANDwidth|BWIDth:INTegration..144[:SENSe]:SEMask:BANDwidth|BWIDth[:RESolution]145[:SENSe]:SEMask:BANDwidth|BWIDth[:RESolution]:AU
TO.................................................................................................145[:SENSe]:SEMask:CARRier:AUTO...................................146[:SENSe]:SEMask:CARRier:CPSD ...................................146[:SENSe]:SEMask:CARRier: POWer................................147[:SENSe]:SEMask:FREQuency:SPAN.............................147[:SENSe]:SEMask:GWLan:MODulation...........................147
121

GSP-930 User Manual[:SENSe]:SEMask:HELP:STATe....................................... 148[:SENSe]:SEMask:NWLan:CHANnel:
BANDwidth|BWIDth............................................................... 148[:SENSe]:SEMask:OFFSet<n>:ADDition:
BANDwidth|BWIDth[:RESolution]?.................................. 148[:SENSe]:SEMask:OFFSet<n>:ADDition:
FREQuency:STARt?.............................................................. 149[:SENSe]:SEMask:OFFSet<n>:ADDition:
FREQuency:STOP?............................................................... 149[:SENSe]:SEMask:OFFSet<n>:ADDition:STARt:ABSolute? .................................................................. 150[:SENSe]:SEMask:OFFSet<n>:ADDition:STARt:
RELative? .................................................................................. 150[:SENSe]:SEMask:OFFSet<n>:ADDition:STATe?...... 150[:SENSe]:SEMask:OFFSet<n>:ADDition:
STOP:ABSolute?.................................................................... 151[:SENSe]:SEMask:OFFSet<n>:ADDition:STOP:
RELative? .................................................................................. 151[:SENSe]:SEMask:OFFSet<n>:BANDwidth|
BWIDth[:RESolution]............................................................. 152[:SENSe]:SEMask:OFFSet<n>:BANDwidth|BWIDth[:RE
Solution]:AUTO....................................................................... 152[:SENSe]:SEMask:OFFSet<n>:FREQuency:STARt... 153[:SENSe]:SEMask:OFFSet<n>:FREQuency:STOP .... 153[:SENSe]:SEMask:OFFSet<n>:STARt:ABSolute........ 153[:SENSe]:SEMask:OFFSet<n>:STARt:RELative ........ 154[:SENSe]:SEMask:OFFSet<n>:STATe ........................... 154[:SENSe]:SEMask:OFFSet<n>:STOP:ABSolute......... 154[:SENSe]:SEMask:OFFSet<n>:STOP:ABSolute:COUPle1[:SENSe]:SEMask:OFFSet<n>:STOP:RELative.......... 155[:SENSe]:SEMask:OFFSet<n>:STOP:RELative:COUPle1[:SENSe]:SEMask:OFFSet<n>:TEST .............................. 156[:SENSe]:SEMask:SELect................................................... 157[:SENSe]:SEMask:TYPE ...................................................... 157[:SENSe]:SEMask:W3GPP:DUPLex:TYPE .................... 157[:SENSe]:SEMask:W3GPP:FDD:ADDition:Limit .......... 158[:SENSe]:SEMask:W3GPP:FDD:ADDition:MOPower 158[:SENSe]:SEMask:W3GPP:FDD:MOPower ................... 159
122

エラー! スタイルが定義されていません。
[:SENSe]:SEMask:W3GPP:FDD:TRANsmit:MODE.....159[:SENSe]:SEMask:W3GPP:TDD:CHIP:RATE................159[:SENSe]:SEMask:W3GPP:TDD:MOPower....................160[:SENSe]:SEMask:W3GPP:TDD:TRANsmit:MODE.....160[:SENSe]:SEMask:WIMax:CHANnel:BANDwidth|BWIDth1[:SENSe]:SEQuence<n>:DELete ......................................161[:SENSe]:SWEep:EGATe:DELay .......................................161[:SENSe]:SWEep:EGATe:LENGth.....................................161[:SENSe]:SWEep:EGATe:STATe ......................................162[:SENSe]:SWEep:TIME..........................................................162[:SENSe]:SWEep:TIME:AUTO ............................................162[:SENSe]:TOI:REFerence ....................................................163[:SENSe]:TOI:LIMit.................................................................163[:SENSe]:ACPR:ACHannel<n>:BANDwidth|B
WIDth

S et
Query

Description

Sets or queries the adjacent channel bandwidth forthe selected adjacent channel. Used with ACPR
measurement.

Syntax[:SENSe]:ACPR:ACHannel<n>:BANDwidth|BWIDth<freq>

Query Syntax[:SENSe]:ACPR:ACHannel<n>:BANDwidth|BWIDth?

Parameter/<freq>
Return parameter
Example<NR3> Hz:ACPR:ACH1:BAND 2.0e+6
S et[:SENSe]:ACPR:ACHannel<n>:HLIMit

Query

Description

Sets or queries the high limit for the selectedadjacent channel. Used with ACPR measurement.

Syntax[:SENSe]:ACPR:ACHannel<n>:HLIMit <ampl>

Query Syntax[:SENSe]:ACPR:ACHannel<n>:HLIMit?
123

GSP-930 User Manual
Parameter/<ampl>
Return parameter
Example<NR3> dBm

:ACPR:ACH1:HLIM -3.0e+1
S et
Query

[:SENSe]:ACPR:ACHannel<n>:LLIMit
Description

Sets or queries the low limit for the selected adjacentchannel. Used with ACPR measurement.

Syntax[:SENSe]:ACPR:ACHannel<n>:LLIMit <ampl>

Query Syntax[:SENSe]:ACPR:ACHannel<n>:LLIMit?

Parameter/<ampl>
Return parameter
Example<NR3> dBm

:ACPR:ACH1:LLIM -5.0e+1
S et
Query

[:SENSe]:ACPR:ACHannel<n>:OFFSet
Description

Sets or queries the adjacent channel offset for theselected adjacent channel. Used with ACPR
measurement.

Syntax[:SENSe]:ACPR:ACHannel<n>:OFFSet <freq>

Query Syntax[:SENSe]:ACPR:ACHannel<n>:OFFSet?

Parameter/<freq>
Return parameter
Example<NR3> Hz:ACPR:ACH1:OFFSet 2.0e+6
S et[:SENSe]:ACPR:BANDwidth|BWIDth

Query

Description

Sets or queries the main channel bandwidth for ACPR
measurements.

Syntax[:SENSe]:ACPR:BANDwidth|BWIDth <freq>

Query Syntax[:SENSe]:ACPR:BANDwidth|BWIDth?

124

エラー! スタイルが定義されていません。
Parameter/<freq>
Return parameter
Example<NR3> Hz:ACPR: BAND 2.0e+6
S et
Query

[:SENSe]:ACPR:HLIMit
Description

Sets or queries the high limit for the main channel.
Used with ACPR measurement.

Syntax[:SENSe]:ACPR:HLIMit <ampl>

Query Syntax[:SENSe]:ACPR:HLIMit?

Parameter/<ampl>
Return parameter
Example<NR3>:ACPR: HLIM -3.0e+1
S et
Query

[:SENSe]:ACPR:LLIMit
Description

Sets or queries the low limit for the main channel.
Used with ACPR measurement.

Syntax[:SENSe]:ACPR:LLIMit <ampl>

Query Syntax[:SENSe]:ACPR:LLIMit?

Parameter/<ampl>
Return parameter
Example<NR3>:ACPR:ACH1:LLIM -5.0e+1
S et[:SENSe]:ACPR:HELP
Description

Turns the on-screen help on/off.

Syntax[:SENSe]:ACPR:HELP {OFF|ON|0|1}

Parameter

01OFFTurn help off.
Turn help on.
Turn help off.
125

GSP-930 User Manual
ON
ExampleTurn help on.

:ACPR:HELP ON
S et
Query

[:SENSe]:ACPR:SPACe
Description

Sets or queries the channel spacing between the main
channels.

Syntax[:SENSe]:ACPR:SPACe <freq>

Query Syntax[:SENSe]:ACPR:SPACe?

Parameter/<freq>
Return parameter
Example<NR3> Hz:ACPR: SPAC 2.0e+6
S et[:SENSe]:ASET:AMPLitude

Query

Description

Sets or queries the autoset amplitude floor level.

Syntax[:SENSe]:ASET:AMPLitude <ampl>

Query Syntax[:SENSe]:ASET:AMPLitude?

Parameter/<ampl>
Return parameter
Example<NR3>:ASET:AMPL 8.0e+1
S et[:SENSe]:ASET:AMPLitude:AUTO

Query

Description

Sets autoset amplitude floor level to auto or manual.

Syntax[:SENSe]:ASET:AMPLitude:AUTO {OFF|ON|0|1}

Query Syntax[:SENSe]:ASET:AMPLitude:AUTO?

Parameter

0
1
OFF
ON

126

Turn autoset amplitude floor to manual.
Turn autoset amplitude floor to auto.
Turn autoset amplitude floor to manual.
Turn autoset amplitude floor to auto.

エラー! スタイルが定義されていません。
Return parameter 0
1
ExampleAutoset amplitude floor is in manual.
Autoset amplitude floor is in auto.

:ASET:AMPL:AUTO 1
S et[:SENSe]:ASET:RUN
Description

Activates the Autoset function.

Syntax[:SENSe]:ASET:RUN

Example:ASET:RUN
S et
Query

[:SENSe]:ASET:SPAN
Description

Sets or queries the Autoset span.

Syntax[:SENSe]:ASET:SPAN <freq>

Query Syntax[:SENSe]:ASET:SPAN?

Parameter/<freq>
Return parameter
Example<NR3> Hz

ASET:SPAN 2.0e+6
S et[:SENSe]:ASET:SPAN:AUTO

Query

Description

Turns the Autoset span to auto or manual.

Syntax[:SENSe]:ASET:SPAN:AUTO {OFF|ON|0|1}

Query Syntax[:SENSe]:ASET:SPAN:AUTO?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurn Autoset span to manual (off).
Turn Autoset span to automatic (on).
Turn Autoset span to manual (off).
Turn Autoset span to automatic (on).
Autoset span is set to manual (off).
Autoset span is set to automatic (on).

:ASET:SPAN:AUTO
127

GSP-930 User Manual
S et
Query

[:SENSe]:AVERage:COUNt
Description

Sets the number of traces that are used with theaverage function.

Syntax[:SENSe]:AVERage:COUNt <integer>

Query Syntax[:SENSe]:AVERage:COUNt?

Parameter/<integer>
Return parameter
Example<NR1>:AVER:COUN 20
S et[:SENSe]:AVERage:STATe

Query

Description

Turns the Average function on/off.

Syntax[:SENSe]:AVERage:STATe {OFF|ON|0|1}

Query Syntax[:SENSe]:AVERage:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurn the Average function off.
Turn the Average function on.
Turn the Average function off.
Turn the Average function on.
The Average function is off.
The Average function is on.

:AVER:STAT ON[:SENSe]:AVERage:TYPE

S et

Description

Sets the method that the Average function uses tocalculate the average.

Syntax[:SENSe]:AVERage:TYPE{VOLTage|LOGarithmic|POWer}

Query syntax[:SENSe]:AVERage:TYPE?

128

エラー! スタイルが定義されていません。
Parameter/Retur VOLTagen parameter
LOGarithmic
POWer
ExampleSets Average to voltage
Sets Average to logarithmic
Sets Average to power:AVER:TYPE VOLT
S et
Query

[:SENSe]:BANDwidth|BWIDth[:RESolution]
Description

Sets the resolution bandwidth (RBW).

Syntax[:SENSe]:BANDwidth|BWIDth[:RESolution] <freq>

Query Syntax[:SENSe]:BANDwidth|BWIDth[:RESolution]?

Parameter/<freq>
Return parameter
Example<NR3> Hz

:BAND 1.0e+6[:SENSe]:BANDwidth|BWIDth[:RESolution]:AUTO

S et
Query

Description

Turns the RBW to auto (on) or manual (off).

Syntax[:SENSe]: BANDwidth|BWIDth[:RESolution]:AUTO{OFF|ON|0|1}

Query Syntax[:SENSe]: BANDwidth|BWIDth[:RESolution]:AUTO?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurn RBW to manual (off).
Turn RBW to automatic (on).
Turn RBW to manual (off).
Turn RBW to automatic (on).
RBW is set to manual (off).
RBW is set to automatic (on).

:BAND:AUTO ON
S et[:SENSe]:BANDwidth|BWIDth:VIDeo

Query

129

GSP-930 User Manual
Description

Sets the video bandwidth (VBW).

Syntax[:SENSe]:BANDwidth|BWIDth:VIDeo <freq>

Query Syntax[:SENSe]:BANDwidth|BWIDth:VIDeo?

Parameter/<freq>
Return parameter
Example<NR3> HzBAND:VID 1.0e+6
S et[:SENSe]:BANDwidth|BWIDth:VIDeo:AUTO

Query

Description

Turns the VBW to auto (on) or manual (off).

Syntax[:SENSe]:BANDwidth|BWIDth:VIDeo:AUTO{OFF|ON|0|1}

Query Syntax[:SENSe]:BANDwidth|BWIDth:VIDeo:AUTO?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurn VBW to manual (off).
Turn VBW to automatic (on).
Turn VBW to manual (off).
Turn VBW to automatic (on).
VBW is set to manual (off).
VBW is set to automatic (on).

:BAND:VID:AUTO OFF[:SENSe]:CHANnel:SPACe:DOWN

S et

Description

Moves to the previous main channel when using
measurements that have a channel space setting.

Syntax[:SENSe]:CHANnel:SPACe:DOWN

Example:CHAN:SPAC:DOWN[:SENSe]:CHANnel:SPACe:UP

130

S et

エラー! スタイルが定義されていません。
Description

Moves to the next main channel when using
measurements that have a channel space setting.

Syntax[:SENSe]:CHANnel:SPACe:UP

Example:CHAN:SPAC:UP
S et
Query

[:SENSe]:CNR:CHANnel:SPACe
Description

Sets the channel space bandwidth for CNR
measurements.

Syntax[:SENSe]:CNR:CHANnel:SPACe <freq>

Query Syntax[:SENSe]:CNR:CHANnel:SPACe?

Parameter/<freq>
Return parameter
Example<NR3> Hz

:CNR:CHAN:SPAC 6.0e+6[:SENSe]:CNR:DELTamarker:MODE

S etDescription

Turns the CNR Noise Marking function to Min(AUTO)
or ΔMarker(MANual).

Syntax[:SENSe]:CNR:DELTamarker:MODE {AUTO|MANual }Parameter

AUTO
ΔMarker

Example:CNR:DELT:MODE AUTO

Sets the Noise Marking to Min.
Sets the Noise Marking to ΔMarker.

[:SENSe]:CORRection:CSET<n>:DATAS et

Description

Sets an offset for a certain frequency for a selected
correction set.

Syntax[:SENSe]:CORRection:CSET<n>:DATA <freq>,<offset>

131

GSP-930 User Manual
Parameter<freq><offset><n><NR3> Hz
<NR3> dB
<NR1>correction set number

ExampleCORR:CSET1:DATA 2e+6,30
S et
Query

[:SENSe]:CORRection:CSET<n>:STATe
Description

Turns the selected correction set on/off.

Syntax[:SENSe]:CORRection:CSET<n>:STATe {OFF|ON|0|1}

Query Syntax[:SENSe]:CORRection:CSET<n>:STATe?

Parameter

0
1
OFF
ON
<n>

Return parameter 0
1
ExampleTurn turn the selected correction set off.
Turn turn the selected correction set on.
Turn turn the selected correction set off.
Turn turn the selected correction set on.
<NR1>correction set number
The selected correction set is off.
The selected correction set is on.

:CORR:CSET1:STAT ON
S et
Query

[:SENSe]:CORRection:CSET<n>:DELete
Description

Deletes the chosen correction set.

Syntax[:SENSe]:CORRection:CSET<n>:DELete

Parameter<n>

Example:CORR:CSET1:DEL 2<NR1>correction set number

S et[:SENSe]:CSO:CHANnel:SPACe

Query

Description

Sets the channel space bandwidth for CSO
measurements.

Syntax[:SENSe]:CSO:CHANnel:SPACe <freq>132エラー! スタイルが定義されていません。
Query Syntax[:SENSe]:CSO:CHANnel:SPACe?

Parameter/<freq>
Return parameter
Example<NR3> Hz

:CSO:CHAN:SPAC 6.0e+6
S et
Query

[:SENSe]:CTB:CHANnel:SPACe
Description

Sets the channel space bandwidth for CTB
measurements.

Syntax[:SENSe]:CTB:CHANnel:SPACe <freq>

Query Syntax[:SENSe]:CTB:CHANnel:SPACe?

Parameter/<freq>
Return parameter
Example<NR3> Hz

:CTB:CHAN:SPAC 6.0e+6[:SENSe]:DEMod:EARPone:TYPE

S et

Description

Sets the demodulation type for the Ear Phone Out
demodulation function.

Syntax[:SENSe]:DEMod:EARPone:TYPE {AM|FM}

ParameterAM
FM

Example:DEM:EARP:TYPE AM

AM demodulation
FM demodulation

S et[:SENSe]:DEMod:EARPone:VOLume

Query

Description

Sets the volume setting for the demodulation
function.

Syntax[:SENSe]:DEMod:EARPone:VOLume <integer>

Query Syntax[:SENSe]:DEMod:EARPone:VOLume?

133

GSP-930 User Manual
Parameter/<integer>
Return parameter
Example<NR1> 0~15

:DEM:EARP:VOL 7
S et[:SENSe]:DEMod:EARPone:GAIN

Query

Description

Sets the gain setting for the demodulation function.

Syntax[:SENSe]:DEMod:EARPone:GAIN <rel_ampl>

Query Syntax[:SENSe]:DEMod:EARPone:GAIN?

Parameter/<rel_ampl>
Return parameter
Example<NR1> 0~18, 6dB steps:DEM:EARP:GAIN 6
S et[:SENSe]:DEMod:FILTer:LPASs

Query

Description

Sets the low pass filter settings for the AM/FM
Analysis function.

Syntax[:SENSe]:DEMod:FILTer:LPASs {LEVel<n>|Bypass}

Query Syntax[:SENSe]:DEMod:FILTer:LPASs?

Parameter/
BypassReturn parameter <n>

Sets the low pass filter to bypass.
<NR1>1~5

The filters 1 to 5 are shown in the table below. The
GSP-930 will automatically detect the signal
frequency.
AM/FM Signal Frequency (Hz)
Selectable bandwidth of LPF (Hz)
<n>=1
<n>=2
<n>=3
<n>=4
<n>=5
≥78,125 156,250
78,12552,08339,063
31,250≥39,063
78,12539,063
26,042
19,531
15,625≥19,531
39,063
19,531
13,021
9,766
7,813≥7,813
15,625
7,813
5,208
3,906
3,125
≥3,906
7,813
3,906
2,604
1,953
1,563
≥1,953
3,906
1,953
1,302
977
781
≥781
1,563
781
521
391
313
≥391
781
391
260
195
156

134

エラー! スタイルが定義されていません。
≥195
≥78
≥39
≥20
≥8

Example

391
156
78
39
16

195
78
39
20
8

130
52
26
13
5

98
39
20
10
4

78
31
16
8
3

:DEM:FILT:LPAS B
S et
Query

[:SENSe]:DEMod:IFBW
Description

Sets or queries the IF bandwidth for the AM/FM
Analysis function.

Syntax[:SENSe]:DEMod:IFBW <freq>

Query Syntax[:SENSe]:DEMod:IFBW?

Parameter/<freq>
Return parameter
Example<NR3>Hz.

DEM:IFBW 3.0e+5
S et[:SENSe]:DETector[:FUNCtion]

Query

Description

Sets/queries the trace detection mode when in
manual mode.

Syntax[:SENSe]:DETector[:FUNCtion]{AVERage|SAMPle|POSitive|NEGative|NORMal}

Query Syntax[:SENSe]:DETector[:FUNCtion]?

AVERage
Parameter/
Return parameter SAMPle
POSitive
NEGative
NORMalExampleSets the detector mode to Average.
Sets the detector mode to Sample.
Sets the detector mode to Peak+.
Sets the detector mode to Peak-.
Sets the detector mode to Normal.

:DET NORM
S et[:SENSe]:DETector[:FUNCtion]:AUTO

Query

135

GSP-930 User Manual
Description

Turns the trace detection mode to auto (on) or
manual (off).

Syntax[:SENSe]:DETector[:FUNCtion]:AUTO {OFF|ON|0|1}

Query Syntax[:SENSe]:DETector[:FUNCtion]:AUTO?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurn the detection mode to manual (off).
Turn the detection mode to auto (on).
Turn the detection mode to manual (off).
Turn the detection mode to auto (on).
The detection mode is set to manual.
The detection mode is set to automatic.

:DET:AUTO ON
S et
Query

[:SENSe]:EMIFilter:STATe
Description

Turns the EMI filter on/off.

Syntax[:SENSe]:EMIFilter:STATe {OFF|ON|0|1}

Query Syntax[:SENSe]:EMIFilter:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurn the EMI filter off.
Turn the EMI filter on.
Turn the EMI filter off.
Turn the EMI filter on.
The EMI filter is off.
The EMI filter is on.

:EMI:STAT 0[:SENSe]:EMIFilter:BANDwidth|BWIDth[:RESolution]

S et

Description

Sets the EMI filter bandwidth (must be set to theexact bandwidth).

Syntax[:SENSe]:EMIFilter:BANDwidth|BWIDth[:RESolution]<freq>

136エラー! スタイルが定義されていません。
Parameter<freq><NR3>Hz {200Hz, 9kHz, 120kHz}

Example:EMIF:BAND 2.0e+2
S et[:SENSe]:FREQuency:CENTer
Description

Sets the center frequency.

Syntax[:SENSe]:FREQuency:CENTer <freq>

Query Syntax[:SENSe]:FREQuency:CENTer?

Parameter/<freq>
Return parameter
ExampleQuery<NR3>Hz:FREQ:CENT 1.0e+9
S et[:SENSe]:FREQuency:CENTer:STEP

Query

Description

Sets the CF Step frequency.

Syntax[:SENSe]:FREQuency:CENTer:STEP <freq>

Query Syntax[:SENSe]:FREQuency:CENTer:STEP?

Parameter/<freq>
Return parameter
Example<NR3>Hz

FREQ:CENT:STEP 1.0e+3
S et[:SENSe]:FREQuency:CENTer:STEP:AUTO

Query

Description

Turns the CF Step frequency setting to auto (on) or
manual (off).

Syntax[:SENSe]:FREQuency:CENTer:STEP:AUTO{OFF|ON|0|1}

Query Syntax[:SENSe]:FREQuency:CENTer:STEP:AUTO?

Parameter

01OFFTurn CF Step to manual (off).
Turn CF Step to auto (on).
Turn CF Step to manual (off).
137

GSP-930 User Manual
ON
Return parameter 0
1
ExampleTurn CF Step to auto (on).
CF Step is set to manual.
CF Step is set to automatic.

:FREQ:CENT:STEP:AUTO OFF
S et
Query

[:SENSe]:FREQuency:OFFSet
Description

Sets the frequency offset settings.

Syntax[:SENSe]:FREQuency:OFFSet <freq>

Query Syntax[:SENSe]:FREQuency:OFFSet?

Parameter/<freq>
Return parameter
Example<NR3>Hz:FREQ:OFFS 1.0e+6
S et[:SENSe]:FREQuency:SPAN
Description

Sets the span settings.

Syntax[:SENSe]:FREQuency:SPAN <freq>

Query Syntax[:SENSe]:FREQuency:SPAN?

Parameter/<freq>
Return parameter
Example<NR3>Hz:FREQ:SPAN 2.0e+9[:SENSe]:FREQuency:SPAN:FULL
Description

Set the span to Full Span.

Syntax[:SENSe]:FREQuency:SPAN:FULL

Example:FREQ:SPAN:FULL[:SENSe]:FREQuency:SPAN:PREVious
138

Query

S et

S et

エラー! スタイルが定義されていません。
Description

Set the span to the previous span setting.

Syntax[:SENSe]:FREQuency:SPAN:PREVious

Example:FREQ:SPAN:PREV
S et
Query

[:SENSe]:FREQuency:STARt
Description

Sets the start frequency.

Syntax[:SENSe]:FREQuency:STARt <freq>

Query Syntax[:SENSe]:FREQuency:STARt?

Parameter/<freq>
Return parameter
Example<NR3>Hz

FREQ:STAR 0
S et
Query

[:SENSe]:FREQuency:STOP
Description

Sets the stop frequency.

Syntax[:SENSe]:FREQuency:STOP <freq>

Query Syntax[:SENSe]:FREQuency:STOP?

Parameter/<freq>
Return parameter
Example<NR3>Hz

FREQ:STOP 1.0e+6
S et[:SENSe]:LIMit<n>:DELete
Description

Deletes the chosen limit line.

Syntax[:SENSe]:LIMit<n>:DELete

Parameter<n>

Example:LIM3:DEL<NR1> limit line number

139

GSP-930 User Manual
S et
Query

[:SENSe]:JITTer:OFFSet:STARt
Description

Sets the start offset for phase jitter measurements.

Syntax[:SENSe]:JITTer:OFFSet:STARt <freq>

Query Syntax[:SENSe]:JITTer:OFFSet:STARt?

Parameter/<freq>
Return parameter
Example<NR3>Hz

JITT:OFFS:STAR 1.0e+7
S et
Query

[:SENSe]:JITTer:OFFSet:STOP
Description

Sets the stop offset for phase jitter measurements.

Syntax[:SENSe]:JITTer:OFFSet:STOP <freq>

Query Syntax[:SENSe]:JITTer:OFFSet:STOP?

Parameter/<freq>
Return parameter
Example<NR3>Hz

JITT:OFFS:STOP 1.5e+7
S et[:SENSe]:NDB:BANDwidth|BWIDth

Query

Description

Sets the NdB amplitude for NdB bandwidth
measurements.

Syntax[:SENSe]:NDB:BANDwidth|BWIDth <rel_amp>

Query Syntax[:SENSe]:NDB:BANDwidth|BWIDth?

Parameter/<rel_ampl>
Return parameter
Example<NR3> dB

:NDB:BAND 3 dB
S et[:SENSe]:OCBW:BANDwidth|BWIDth
140

Query

エラー! スタイルが定義されていません。
Description

Sets the OCBW bandwidth for OCBW measurements.

Syntax[:SENSe]:OCBW:BANDwidth|BWIDth <freq>

Query Syntax[:SENSe]:OCBW:BANDwidth|BWIDth?

Parameter/<freq>
Return parameter
Example<NR3>Hz:OCBW:BAND 4.5+6
S et
Query

[:SENSe]:OCBW:PERCent
Description

Sets or queries the OCBW percentage (OCBW %)parameter.

Syntax[:SENSe]:OCBW:PERCent <integer>

Query Syntax[:SENSe]:OCBW:PERCent?

Parameter/<integer>
Return parameter
Example<NR1>0~100:OCBW:PERC 90
S et
Query

[:SENSe]:OCBW:SPACe
Description

Sets the OCBW channel space for OCBW
measurements.

Syntax[:SENSe]:OCBW:SPACe <freq>

Query Syntax[:SENSe]:OCBW:SPACe?

Parameter/<freq>
Return parameter
Example<NR3>Hz:OCBW:SPAC 6e+7
S et[:SENSe]:OCBW:SPAN

Query

Description

Sets the OCBW span for OCBW measurements.

Syntax[:SENSe]:OCBW:SPAN <freq>
141

GSP-930 User Manual
Query Syntax[:SENSe]:OCBW:SPAN?

Parameter/<freq>
Return parameter
Example<NR3>Hz:OCBW:SPAN 1e+7
S et[:SENSe]:PMETer:FREQuency

Query

Description

Sets the power meter measurement frequency.

Syntax[:SENSe]:PMETer:FREQuency <freq>

Query Syntax[:SENSe]:PMETer:FREQuency?

Parameter/<freq>
Return parameter
Example<NR3>Hz:PMET:FREQ 2e+7
S et
Query

[:SENSe]:PMETer:HLIMit
Description

Sets the power meter high limit for pass/fail tests.

Syntax[:SENSe]:PMETer:HLIMit <ampl>

Query Syntax[:SENSe]:PMETer:HLIMit?

Parameter/<ampl>
Return parameter
Example<NR3> dBm

:PMET:HLIM 10
S et[:SENSe]:PMETer:HOLD:STATe

Query

Description

Turns the power meter Max/Min Hold function on/off.

Syntax[:SENSe]:PMETer:HOLD:STATe {OFF|ON|0|1}

Query Syntax[:SENSe]:PMETer:HOLD:STATe?

Parameter

0
1
OFF

142

Turn the Max/Min Hold function off.
Turn the Max/Min Hold function on.
Turn the Max/Min Hold function off.

エラー! スタイルが定義されていません。
ON
Return parameter 0
1
ExampleTurn the Max/Min Hold function on.
The Max/Min Hold function is off.
The Max/Min Hold function is on.

:PEMT:HOLD:STAT 0
S et
Query

[:SENSe]:PMETer:LLIMit
Description

Sets the power meter low limit for pass/fail tests.

Syntax[:SENSe]:PMETer:LLIMit <ampl>

Query Syntax[:SENSe]:PMETer:LLIMit?

Parameter/<ampl>
Return parameter
Example<NR3> dBm

:PMET:LLIM 0
S et[:SENSe]:PMETer:PSENsor:MODE

Query

Description

Sets or queries the power meter sensor mode.

Syntax[:SENSe]:PMETer:PSENsor:MODE{LOWNoise|FASTer}

Query Syntax[:SENSe]:PMETer:PSENsor:MODE?

Parameter/
LOWNoise
Return parameter FASTer
ExampleSets the power meter mode to low noise.
Set the power meter mode to fast.

:PMET:PSEN:MODE
S et[:SENSe]:PMETer:RECording:TIME

Query

Description

Sets or queries the power meter recording time.

Syntax[:SENSe]:PMETer:RECording:TIME<hour>,<minute>,<second>

Query Syntax[:SENSe]:PMETer:RECording:TIME?

143

GSP-930 User Manual
Parameter/<hour>
Return parameter <minute><second>
Example<NR1>Recording time: hours.
<NR1>Recording time: minutes.
<NR1>Recording time: seconds.

:PMET:REC:TIME 1,10,30
S et[:SENSe]:PMETer:RECording:TIME:STEP

Query

Description

Sets or queries the power meter recording interval inseconds.

Syntax[:SENSe]:PMETer:RECording:TIME:STEP <time>

Query Syntax[:SENSe]:PMETer:RECording:TIME:STEP?

Parameter/<time>
Return parameter
Example<NR1>time in seconds.

:PMET:REC:TIME:STEP 10s
S et[:SENSe]:POWer[:RF]:GAIN

Query

Description

Sets the preamplifier to Auto or Bypass mode.

Syntax[:SENSe]:POWer[:RF]:GAIN {AUTO|BYPASS}

Query Syntax[:SENSe]:POWer[:RF]:GAIN?

Parameter/
AUTO
Return parameter BYPASSExampleSets the preamplifier to auto mode.
Sets the preamplifier to bypass mode.

:POW:GAIN AUTO[:SENSe]:SEMask:BANDwidth|BWIDth:
INTegration

S et
Query

Description

Sets the channel integration bandwidth for SEM
measurements (user defined only).

Syntax[:SENSe]:SEMask:BANDwidth|BWIDth:INTegration<freq>

144

エラー! スタイルが定義されていません。
Query Syntax[:SENSe]:SEMask:BANDwidth|BWIDth:INTegration?

Parameter/<freq>
Return parameter
Example<NR3>Hz:SEM:BAND:INT 3.84e+6[:SENSe]:SEMask:BANDwidth|BWIDth[:RESolution]

S et
Query

Description

Sets the RBW for SEM measurements.

Syntax[:SENSe]:SEMask:BANDwidth|BWIDth[:RESolution]<freq>

Query Syntax[:SENSe]:SEMask:BANDwidth|BWIDth[:RESolution]?

Parameter/<freq>
Return parameter
Example<NR3>Hz:SEM:BAND 2.2e+4[:SENSe]:SEMask:BANDwidth|BWIDth[:RES
olution]:AUTO

S et
Query

Description

Turns the RBW setting to auto (on) or manual (off) for
SEM measurements.

Syntax[:SENSe]:SEMask:BANDwidth|BWIDth[:RESolution]:
AUTO {OFF|ON|0|1}

Query Syntax[:SENSe]:SEMask:BANDwidth|BWIDth[:RESolution]:
AUTO?

Parameter

0
1
OFF
ON

Return parameter 0
1Turn RBW to manual (off).
Turn RBW to auto (on).
Turn RBW to manual (off).
Turn RBW to auto (on).
RBW is set to manual.
RBW is set to automatic.

145

GSP-930 User Manual
Example:SEM:BAND: AUTO OFF
S et[:SENSe]:SEMask:CARRier:AUTO

Query

Description

Turns the PSDRef or TotalPwrRef modes to auto (on)or manual (off) for SEM measurements.

Syntax[:SENSe]:SEMask:CARRier:AUTO {OFF|ON|0|1}

Query Syntax[:SENSe]:SEMask:CARRier:AUTO?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurn PSDRef/TotalPwrRef to manual(off).
Turn PSDRef/TotalPwrRef to auto (on).
Turn PSDRef/TotalPwrRef to manual(off).
Turn PSDRef/TotalPwrRef to auto (on).
PSDRef/TotalPwrRef is set to manual.
PSDRef/TotalPwrRef is set toautomatic.

:SEM:CARR:AUTO OFF
S et[:SENSe]:SEMask:CARRier:CPSD

Query

Description

Sets the PSDRef for SEM measurements.

Syntax[:SENSe]:SEMask:CARRier:CPSD <NR3>

Query Syntax[:SENSe]:SEMask:CARRier:CPSD?

Parameter/<NR3>
Return parameter
Example146

PSD ref unit = dBm/Hz:SEM:CARR:CPSD 20

エラー! スタイルが定義されていません。
S et
Query

[:SENSe]:SEMask:CARRier: POWer
Description

Sets the TotalPwrRef amplitude for SEM
measurements.

Syntax[:SENSe]:SEMask:CARRier: POWer <ampl>

Query Syntax[:SENSe]:SEMask:CARRier: POWer?

Parameter/<ampl>
Return parameter
Example<NR3>:SEM:CARR:POW 2 dbm
S et
Query

[:SENSe]:SEMask:FREQuency:SPAN
Description

Sets the channel span for SEM measurements (userdefined only).

Syntax[:SENSe]:SEMask:FREQuency:SPAN<freq>

Query Syntax[:SENSe]:SEMask:FREQuency:SPAN?

Parameter/<freq>
Return parameter
Example<NR3>Hz:SEM:FREQ:SPAN 2.2e+7
S et[:SENSe]:SEMask:GWLan:MODulation

Query

Description

Sets the modulation type for the 802.11g SEM
measurement.

Syntax[:SENSe]:SEMask:GWLan:MODulation{SINGle|MULTIcarrier}

Query Syntax[:SENSe]:SEMask:GWLan:MODulation?

Parameter/
SINGleReturn parameter MULTIcarrier
ExampleERP-OFDM/DSSS-OFDN
ERP-DSSS/ERP-PBCC/ERP-CCK:SEM:GWL:MOD SING
147

GSP-930 User Manual
S et
Query

[:SENSe]:SEMask:HELP:STATe
Description

Turns the on-screen help window on/off.

Syntax[:SENSe]:SEMask:HELP:STATe {OFF|ON|0|1}

Query Syntax[:SENSe]:SEMask:HELP:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurns the help window off.
Turns the help window on.
Turns the help window off.
Turns the help window on.
Help window is off.
Help window is on.

:SEM:HELP:STATE 1[:SENSe]:SEMask:NWLan:CHANnel:
BANDwidth|BWIDth

S et
Query

Description

Sets the modulation type for the 802.11g SEM
measurement. Only 20MHz or 40MHz can be used.

Syntax[:SENSe]:SEMask:NWLan:CHANnel:
BANDwidth|BWIDth <freq>

Parameter/<freq>
Return parameter
Example<NR3> 20 MHz or 40MHz (2e+7 |4e+7)can be selected.

:SEM:NWL:CHAN:BAND 20 MHZ[:SENSe]:SEMask:OFFSet<n>:ADDition:
BANDwidth|BWIDth[:RESolution]?
Description

148

Query

Returns the RBW of the selected offset for the
additional requirements of the selected 3GPP SEM
test.

エラー! スタイルが定義されていません。
Query syntax[:SENSe]:SEMask:OFFSet<n>:ADDition:
BANDwidth|BWIDth[:RESolution]?

Parameter/<n>Return parameter <NR3>
ExampleOffset 1 to 5
RBW in Hz:SEM:OFFS1:ADD:BAND?
>XXXXXXXXXXXx[:SENSe]:SEMask:OFFSet<n>:ADDition:
FREQuency:STARt?

Query

Description

Returns the start frequency of the selected offset for
the additional requirements of the selected 3GPP
SEM test.

Query syntax[:SENSe]:SEMask:OFFSet<n>:ADDition:
FREQuency:STARt?

Parameter/<n>Return parameter <NR3>
ExampleOffset 1 to 5
Start frequency in Hz:SEM:OFFS1:ADD:FREQ:STAR?
>2.5e+6[:SENSe]:SEMask:OFFSet<n>:ADDition:
FREQuency:STOP?

Query

Description

Returns the stop frequency of the selected offset for
the additional requirements of the selected 3GPP
SEM test.

Query syntax[:SENSe]:SEMask:OFFSet<n>:ADDition:
FREQuency:STOP?

Parameter/<n>Return parameter <NR3>
ExampleOffset 1 to 5
Stop frequency in Hz:SEM:OFFS1:ADD:FREQ:STOP?
>3.5e+6149

GSP-930 User Manual[:SENSe]:SEMask:OFFSet<n>:ADDition:STARt:ABSolute?

Query

Description

Returns the “start” amplitude (dBm) of the Absolute
Mask for the selected offset for the additional
requirements of the selected 3GPP SEM test.

Query syntax[:SENSe]:SEMask:OFFSet<n>:ADDition:STARt:
ABSolute?

Parameter/<n>Return parameter <NR3>
ExampleOffset 1 to 5
Amplitude at start frequency:SEM:OFFS1:ADD:STAR:ABS?
>1.5e+1[:SENSe]:SEMask:OFFSet<n>:ADDition:STARt: RELative?

Query

Description

Returns the “start” amplitude (dBc) of the Relative
Mask for the selected offset for the additional
requirements of the selected 3GPP SEM test.

Query syntax[:SENSe]:SEMask:OFFSet<n>:ADDition:STARt:
RELative?

Parameter/<n>Return parameter <NR3>
ExampleOffset 1 to 5
Relative amplitude at startfrequency:SEM:OFFS1:ADD:STAR:REL?
>1.5e+1[:SENSe]:SEMask:OFFSet<n>:ADDition:
STATe?
Description

150

Query

Returns the state mask for the additional
requirements for the selected offset.

エラー! スタイルが定義されていません。
Syntax[:SENSe]:SEMask:OFFSet<n>:ADDition:STATe?

Return parameter 0
1
ExampleMask is off.
Mask is on.

:SEM:OFF1:ADD:STAT?
>0[:SENSe]:SEMask:OFFSet<n>:ADDition:
STOP:ABSolute?

Query

Description

Returns the “Stop” amplitude (dBm) of the Absolute
Mask for the selected offset for the additional
requirements of the selected 3GPP SEM test.

Query syntax[:SENSe]:SEMask:OFFSet<n>:ADDition:STOP:
ABSolute?

Parameter/<n>Return parameter <NR3>
ExampleOffset 1 to 5
Amplitude at stop frequency:SEM:OFFS1:ADD:STOP:ABS?
>1.5e+1[:SENSe]:SEMask:OFFSet<n>:ADDition:
STOP: RELative?

Query

Description

Returns the “stop” amplitude (dBc) of the Relative
Mask for the selected offset for the additional
requirements of the selected 3GPP SEM test.

Query syntax[:SENSe]:SEMask:OFFSet<n>:ADDition:STOP:
RELative?

Parameter/<n>Return parameter <NR3>
ExampleOffset 1 to 5
Relative amplitude at stopfrequency:SEM:OFFS1:ADD:STOP:REL?
>1.5e+1151

GSP-930 User Manual[:SENSe]:SEMask:OFFSet<n>:BANDwidth|
BWIDth[:RESolution]

S et
Query

Description

Sets or queries the resolution bandwidth of theselected offset.

Syntax[:SENSe]:SEMask:OFFSet<n>:BANDwidth|BWIDth[:RESolution] <freq>

Query Syntax[:SENSe]:SEMask:OFFSet<n>:BANDwidth|BWIDth[:RESolution]?

Parameter/<freq>
Return parameter <n>Example<NR3> Hz
<NR1>offset 1~5

SEM:OFFS1:BAND 3.0e+3[:SENSe]:SEMask:OFFSet<n>:BANDwidth|B
WIDth[:RESolution]:AUTO

S et
Query

Description

Turns the resolution bandwidth for the selectedchannel to manual or automatic mode.

Syntax[:SENSe]:SEMask:OFFSet<n>:BANDwidth|BWIDth[:RESolution]:AUTO {OFF|ON|0|1}

Query Syntax[:SENSe]:SEMask:OFFSet<n>:BANDwidth|BWIDth[:RESolution]:AUTO?

Parameter

0
1
OFF
ON

Return parameter 0
1

152

Set RBW to manual.
Set RBW to auto.
Set RBW to manual.
Set RBW to auto.
RBW is set to manual.
RBW is set to auto.

エラー! スタイルが定義されていません。

[:SENSe]:SEMask:OFFSet<n>:FREQuency:
STARt

S et
Query

Description

Sets or queries the start frequency of the selected
offset.

Syntax[:SENSe]:SEMask:OFFSet<n>:FREQuency:STARt<freq>

Query Syntax[:SENSe]:SEMask:OFFSet<n>:FREQuency:STARt?

Parameter/<freq>
Return parameter <n>Example<NR3> Hz
<NR1>offset 1~5

SEM:OFFS1:FREQ:STAR 2.5e+3[:SENSe]:SEMask:OFFSet<n>:FREQuency:
STOP

S et
Query

Description

Sets or queries the stop frequency of the selected
offset.

Syntax[:SENSe]:SEMask:OFFSet<n>:FREQuency:STOP<freq>

Query Syntax[:SENSe]:SEMask:OFFSet<n>:FREQuency:STOP?

Parameter/<freq>
Return parameter <n>Example<NR3> Hz
<NR1>offset 1~5

SEM:OFFS1:FREQ:STOP 2.5e+3[:SENSe]:SEMask:OFFSet<n>:STARt:
ABSolute

S et
Query

Description

Sets or queries the amplitude of the start frequencyof the Absolute Mask for the selected offset.

Syntax[:SENSe]:SEMask:OFFSet<n>:STARt:ABSolute <ampl>

Query Syntax[:SENSe]:SEMask:OFFSet<n>:STARt:ABSolute?
153

GSP-930 User Manual
Parameter/<ampl>
Return parameter <n>Example<NR3> dBm
<NR1>offset 1~5

SEM:OFFS1:STAR:ABS 1.5e+1[:SENSe]:SEMask:OFFSet<n>:STARt:
RELative

S et
Query

Description

Sets or queries the amplitude of the start frequencyof the Relative Mask for the selected offset.

Syntax[:SENSe]:SEMask:OFFSet<n>:STARt:RELative <ampl>

Query Syntax[:SENSe]:SEMask:OFFSet<n>:STARt:RELative?

Parameter/<ampl>
Return parameter <n>Example<NR3> dBc
<NR1> offset 1~5

SEM:OFFS1:STAR:REL 2.5e+1
S et
Query

[:SENSe]:SEMask:OFFSet<n>:STATe
Description

Turns the selected offset on/off.

Syntax[:SENSe]:SEMask:OFFSet<n>:STATe {OFF|ON|0|1}

Query Syntax[:SENSe]:SEMask:OFFSet<n>:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurns the selected offset off.
Turns the selected offset on.
Turns the selected offset off.
Turns the selected offset on.
The selected offset is off.
The selected offset is on.

SEM:OFFS1:STAT 1[:SENSe]:SEMask:OFFSet<n>:STOP:ABSolute

154

S et
Query

エラー! スタイルが定義されていません。
Description

Sets or queries the amplitude of the stop frequencyof the Absolute Mask for the selected offset.

Syntax[:SENSe]:SEMask:OFFSet<n>:STOP:ABSolute <ampl>

Query Syntax[:SENSe]:SEMask:OFFSet<n>:STOP:ABSolute?

Parameter/<ampl>
Return parameter <n>Example<NR3> dBm
<NR1>offset 1~5

SEM:OFFS1:STOP:ABS 1.5e+1[:SENSe]:SEMask:OFFSet<n>:STOP:
ABSolute:COUPle

S et
Query

Description

Couples the Absolute Stop amplitude to the Absolute
Start amplitude for the selected offset.

Syntax[:SENSe]:SEMask:OFFSet<n>:STOP:ABSolute:COUPle{OFF|ON|0|1}

Query Syntax[:SENSe]:SEMask:OFFSet<n>:STOP:ABSolute:
COUPle?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurns coupling off.
Turns coupling on.
Turns coupling off.
Turns coupling on.
Coupling is off.
Coupling is on.

SEM:OFFS1:STOP:ABS:COUP 0[:SENSe]:SEMask:OFFSet<n>:STOP:
RELative

S et
Query

Description

Sets or queries the amplitude of the stop frequencyof the Relative Mask for the selected offset.

Syntax[:SENSe]:SEMask:OFFSet<n>:STOP:RELative <ampl>

155

GSP-930 User Manual
Query Syntax[:SENSe]:SEMask:OFFSet<n>:STOP:RELative?

Parameter/<ampl>
Return parameter <n>Example<NR3> dBc
<NR1>offset 1~5

SEM:OFFS1:STOP:REL 1.5e+1[:SENSe]:SEMask:OFFSet<n>:STOP:
RELative:COUPle

S et
Query

Description

Couples the Relative Stop amplitude to the Relative
Start amplitude for the selected offset.

Syntax[:SENSe]:SEMask:OFFSet<n>:STOP:RELative:COUPle{OFF|ON|0|1}

Query Syntax[:SENSe]:SEMask:OFFSet<n>:STOP:RELative:
COUPle?

Parameter

0
1
OFF
ON

Return parameter 0
1

Turns coupling off.
Turns coupling on.
Turns coupling off.
Turns coupling on.
Coupling is off.
Coupling is on.
S et[:SENSe]:SEMask:OFFSet<n>:TEST

Query

Description

Sets or queries the masks to use for the Fail Mask(s).

Syntax[:SENSe]:SEMask:OFFSet<n>:TEST{ABS|REL|AND|OR}

Query Syntax[:SENSe]:SEMask:OFFSet<n>:TEST?

ABS
Parameter/
Return parameter REL
AND
ORExample156

Absolute mask
Relative mask
Absolute and Relative mask
Absolute or Relative mask

:SEM:OFFS:1:TEST ABS

エラー! スタイルが定義されていません。
S et
Query

[:SENSe]:SEMask:SELect
Description

Selects or queries the type of spectrum emission
mask.

Syntax[:SENSe]:SEMask:SELect{MANual|W3GPP|BWLan|GWLan|NWLan|WIMax}

Query Syntax[:SENSe]:SEMask:SELect?

MANualParameter/
Return parameter W3GPP
BWLan
GWLan
NWLan
WIMax
ExampleUser-defined SEM
3GPP SEM
802.11b SEM
802.11g SEM
802.11n SEM
802.16 SEM:SEM:SEL MAN
S et
Query

[:SENSe]:SEMask:TYPE
Description

Selects or queries the method used as the reference
for calculating the offset power: Total powerreference or power spectral density reference.

Syntax[:SENSe]:SEMask:TYPE {PSDRef|TPRef}

Query Syntax[:SENSe]:SEMask:TYPE?

Parameter/
PSDRefReturn parameter TPRef
ExamplePower Spectral Density Reference
Total Power Reference:SEM:TYPE PSDR
S et[:SENSe]:SEMask:W3GPP:DUPLex:TYPE

Query

Description

Selects or queries the type of duplexing used for
3GPP tests.

Syntax[:SENSe]:SEMask:W3GPP:DUPLex:TYPE {FDD|TDD}

157

GSP-930 User Manual
Query Syntax[:SENSe]:SEMask:W3GPP:DUPLex:TYPE?

Parameter/
FDD
Return parameter TDD
ExampleFrequency-division duplexing
Time-division duplexing:SEM:W3GPP:DUPL:TYPE FDD[:SENSe]:SEMask:W3GPP:FDD:ADDition:
Limit

S et
Query

Description

Selects or queries the operating band used for the
3GPP FDD additional requirements. See the user
manual for a list of the 3GPP operation bands.

Syntax[:SENSe]:SEMask:W3GPP:FDD:ADDition:Limit {NONE |BAND<n>}Query Syntax[:SENSe]:SEMask:W3GPP:FDD:ADDition:Limit?

Parameter/
NONE
Return parameter BAND<n>ExampleWhen n = band number:SEM:W3GPP:FDD:ADD:L BAND4[:SENSe]:SEMask:W3GPP:FDD:ADDition:
MOPower

S et
Query

Description

Selects or queries Max Out Power for the 3GPP
additional requirements for the selected offset. Please
see the user manual for a list of the selectablemaximum power output levels.

Syntax[:SENSe]:SEMask:W3GPP:FDD:ADDition:MOPower{NONE | LEVel<n>}Query Syntax[:SENSe]:SEMask:W3GPP:FDD:ADDition:MOPower?

Parameter/
NONE
Return parameter LEVEL<n>Example158

N= maximum power output level.

:SEM:W3GPP:FDD:ADD:MOP LEV43

エラー! スタイルが定義されていません。
S et[:SENSe]:SEMask:W3GPP:FDD:MOPowerQuery

Description

Selects or queries Max Out Power for the selected
offset. Please see the user manual for a list of theselectable maximum power output levels.

Syntax[:SENSe]:SEMask:W3GPP:FDD:MOPower{NONE | LEVel<n>}Query Syntax[:SENSe]:SEMask:W3GPP:FDD:MOPower?

Parameter/
NONE
Return parameter LEVEL<n>ExampleN= maximum power output level.

:SEM:W3GPP:FDD:MOP LEV43[:SENSe]:SEMask:W3GPP:FDD:TRANsmit:M
ODE

S et
Query

Description

Selects or queries the transmit mode of the FDD
3GPP test: Base station, or User Equipment.

Syntax[:SENSe]:SEMask:W3GPP:FDD:TRANsmit:MODE{BS|UE}

Query Syntax[:SENSe]:SEMask:W3GPP:FDD:TRANsmit:MODE?

Parameter/
BS
Return parameter UE
ExampleBase station
User Equipment:SEM:W3GPP:FDD:TRAN:MODE UE
S et[:SENSe]:SEMask:W3GPP:TDD:CHIP:RATE

Query

Description

Selects or queries the chip rate for TDD 3GPP tests.

Syntax[:SENSe]:SEMask:W3GPP:TDD:CHIP:RATE{3.84e+6|1.28e+6|7.68e+6}

Query Syntax[:SENSe]:SEMask:W3GPP:TDD:CHIP:RATE?

159

GSP-930 User Manual
Parameter/
3.84e+6
Return parameter 1.28e+6
7.68e+6
Example<freq><freq><freq>:SEM:W3GPP:TDD:CHIP:RATE 3.84e+6
S et[:SENSe]:SEMask:W3GPP:TDD:MOPowerQuery

Description

Selects or queries Max Out Power for TDD 3GPP
tests. See the user manual for the a list of the power
levels.

Syntax[:SENSe]:SEMask:W3GPP:TDD:ADDition:MOPower{NONE | LEVel<n>}Query Syntax[:SENSe]:SEMask:W3GPP:TDD:ADDition:MOPower?

Parameter/
LEVEL<n>Return parameter
ExampleN=maximum output power:SEM:W3GPP:TDD:MOP LEV1[:SENSe]:SEMask:W3GPP:TDD:TRANsmit:M
ODE

S et
Query

Description

Selects or queries the transmit mode of the TDD
3GPP test: Base station, or User Equipment.

Syntax[:SENSe]:SEMask:W3GPP:TDD:TRANsmit:MODE{BS|UE}

Query Syntax[:SENSe]:SEMask:W3GPP:TDD:TRANsmit:MODE?

Parameter/
BS
Return parameter UE
ExampleBase station
User Equipment:SEM:W3GPP:TDD:TRAN:MODE UE[:SENSe]:SEMask:WIMax:CHANnel:
BANDwidth|BWIDth
160

S et
Query

エラー! スタイルが定義されていません。
Description

Selects or queries the 802.16 channel bandwidth (10Mor 20M channelization).

Syntax[:SENSe]:SEMask:WIMax:CHANnel:BANDwidth|BWIDth{1e+7|2e+7}

Query Syntax[:SENSe]:SEMask:WIMax:CHANnel:BANDwidth|
BWIDth?

Parameter/
1e+7
Return parameter 2e+7
Example<freq><freq>:SEM:WIM:CHAN:BAND 1e+7[:SENSe]:SEQuence<n>:DELete
Description

Deletes the chosen sequence.

Syntax[:SENSe]:SEQuence<n>:DELete

Parameter<n>

Example:SEQ1:DEL

S et<NR1> sequence 1 to 5.

S et
Query

[:SENSe]:SWEep:EGATe:DELay
Description

Sets or queries the gate delay time.

Syntax[:SENSe]:SWEep:EGATe:DELlay <time>

Query Syntax[:SENSe]:SWEep:EGATe:DELlay?

Parameter/<time>
Return parameter
ExampleGate delay time in seconds

SWE:EGAT:DEL 10 ms
S et[:SENSe]:SWEep:EGATe:LENGth

Query

Description

Sets or queries the gate length time.

Syntax[:SENSe]:SWEep:EGATe:LENGth <time>161

GSP-930 User Manual
Query Syntax[:SENSe]:SWEep:EGATe:LENgth?

Parameter/<time>
Return parameter
ExampleGate length time in seconds

SWE:EGAT:LENG 10 ms
S et
Query

[:SENSe]:SWEep:EGATe:STATe
Description

Turns the gated sweep mode on/off.

Syntax[:SENSe]:SWEep:EGATe:STATe {OFF|ON|0|1}

Query Syntax[:SENSe]:SWEep:EGATe:STATe?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurns gated sweep mode off.
Turns gated sweep mode on.
Turns gated sweep mode off.
Turns gated sweep mode on.
Gated sweep mode is off.
Gated sweep mode is on.

:SWE:EGAT:STAT 1
S et
Query

[:SENSe]:SWEep:TIME
Description

Sets the sweep time.

Syntax[:SENSe]:SWEep:TIME <time>

Query Syntax[:SENSe]:SWEep:TIME?

Parameter/<time>
Return parameter
ExampleSweep time in seconds

SWE:TIME 60 ms
S et[:SENSe]:SWEep:TIME:AUTO
Description

162

Query

Turns the Sweep time setting to auto (on) or manual(off).

エラー! スタイルが定義されていません。
Syntax[:SENSe]:SWEep:TIME:AUTO {OFF|ON|0|1}

Query Syntax[:SENSe]:SWEep:TIME:AUTO?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurn sweep time to manual (off).
Turn sweep time to auto (on).
Turn sweep time to manual (off).
Turn sweep time to auto (on).
Sweep time is set to manual.
Sweep time is set to automatic.

:SWE:TIME:AUTO 0
S et
Query

[:SENSe]:TOI:REFerence
Description

Sets the TOI reference to the upper or lower base.

Syntax[:SENSe]:TOI:REFerence {UPPer|LOWer}

Query Syntax[:SENSe]:TOI:REFerence?

Parameter/
UPPer
Return parameter LOWer
ExampleUpper base.
Lower base.

:TOI:REF UPP
S et
Query

[:SENSe]:TOI:LIMit
Description

Sets the TOI pass/fail limit amplitude.

Syntax[:SENSe]:TOI:LIMit <ampl>

Query Syntax[:SENSe]:TOI:LIMit?

Parameter/<ampl>
Return parameter
ExampledBm

TOI:LIM 30 dbm

SEQuence Commands:SEQuence:MODE ..................................................................164
163

GSP-930 User Manual:SEQuence:TEST:ACTive.................................................... 164
S et
Query

:SEQuence:MODE
Description

Sets the sequence run mode to single or continuous.

Syntax:SEQuence:MODE {SINGle|CONTinuous}

Query Syntax:SEQuence:MODE?

Parameter/
SINGleReturn parameter CONTinuous
ExampleSingle run mode
Continuous run mode

:SEQ:MODE SING

:SEQuence:TEST:ACTive
Description

Runs the current sequence.

Syntax:SEQuence:TEST:ACTive

Example:SEQ1:TEST:ACTS etSOURce Commands
SOURce Commands ............................................................. 164:SOURce:POWer[:LEVel][:IMMediate][:AMPLitude]. 165:SOURce:POWer[:LEVel][:IMMediate][:AMPLitude]:OFFSet...................................................................................... 165:SOURce:POWer[:LEVel][:IMMediate][:AMPLitude]:STEP.......................................................................................... 165:SOURce:POWer[:LEVel][:IMMediate][:AMPLitude]:STEP:AUTO ............................................................................. 166:SOURce:POWer:MODE....................................................... 166:SOURce:POWer:SWEep...................................................... 166

164エラー! スタイルが定義されていません。

:SOURce:POWer[:LEVel][:IMMediate][:AMPLitude]

S et
Query

Description

Sets the tracking generator power level.

Syntax:SOURce:POWer[:LEVel][:IMMediate][:AMPLitude]<ampl>

Query Syntax:SOURce:POWer[:LEVel][:IMMediate][:AMPLitude]?

Parameter/<ampl>
Return parameter
ExampledBm:SOUR:POW 30 dbm:SOURce:POWer[:LEVel][:IMMediate][:AMPLitude]:OFFSet

S et
Query

Description

Sets the tracking generator offset level.

Syntax:SOURce:POWer[:LEVel][:IMMediate][:AMPLitude]:OFFSet <rel_ampl>

Query Syntax:SOURce:POWer[:LEVel][:IMMediate][:AMPLitude]:OFFSet?

Parameter/<rel_ampl>
Return parameter
ExampledB:SOUR:POW:OFFS 10 db:SOURce:POWer[:LEVel][:IMMediate][:AMPLitude]:STEP

S et
Query

Description

Sets the tracking generator step level.

Syntax:SOURce:POWer[:LEVel][:IMMediate][:AMPLitude]:
STEP <rel_ampl>

Query Syntax:SOURce:POWer[:LEVel][:IMMediate][:AMPLitude]:
STEP?
165

GSP-930 User Manual
Parameter/<rel_ampl>
Return parameter
ExampledB:SOUR:POW:STEP .5 db:SOURce:POWer[:LEVel][:IMMediate][:AMPLitude]:STEP:AUTO

S et
Query

Description

Turns the tracking generator step level setting to
auto (on) or manual (off).

Syntax:SOURce:POWer[:LEVel][:IMMediate][:AMPLitude]:STEP:AUTO {OFF|ON|0|1}

Query Syntax:SOURce:POWer[:LEVel][:IMMediate][:AMPLitude]:STEP:AUTO?

Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurn TG step level to manual (off).
Turn TG step level to auto (on).
Turn TG step level to manual (off).
Turn TG step level to auto (on).
TG step level is set to manual.
TG step level is set to automatic.

:SOUR:POW:STEP:AUTO 1
S et
Query

:SOURce:POWer:MODE
Description

Sets the Power Sweep mode.

Syntax:SOURce:POWer:MODE {FIXed|SWEep}

Query Syntax:SOURce:POWer:MODE?

Parameter/
FIXed
Return parameter SWEep
ExamplePower sweep off.
Power sweep on.

:SOUR:POW:MODE FIX
S et:SOURce:POWer:SWEep
166

Query

エラー! スタイルが定義されていません。
Description

Sets the Power Sweep offset level.

Syntax:SOURce:POWer:SWEep <rel_ampl>

Query Syntax:SOURce:POWer:SWEep?

Parameter/<rel_ampl>
Return parameter
ExampledB:SOUR:POW:SWE 10 db

167

GSP-930 User Manual

SYSTem Commands:SYSTem:ALARm:STATe .................................................... 168:SYSTem:CLOCk<n>:DATE................................................ 169:SYSTem:CLOCk<n>:MODE............................................... 169:SYSTem:CLOCk<n>:STATe ............................................. 170:SYSTem:CLOCk<n>:TIME ................................................. 170:SYSTem:COMMunicate:GPIB[:SELF]:ADDRess....... 171:SYSTem:COMMunicate:LAN:ADDRess........................ 171:SYSTem:COMMunicate:LAN:MASK............................... 171:SYSTem:COMMunicate:LAN:GATEway ....................... 171:SYSTem:COMMunicate:LAN:CONFigure..................... 172:SYSTem:COMMunicate:LCI.............................................. 172:SYSTem:COMMunicate:SERial[:RECeive]:BAUD..... 172:SYSTem:COMMunicate:USB:MODE.............................. 172:SYSTem:DATE....................................................................... 173:SYSTem:ERRor:CLEar ........................................................ 173:SYSTem:ERRor[:NEXT]?.................................................... 173:SYSTem:KLOCk .................................................................... 173:SYSTem:PRESet................................................................... 174:SYSTem:PRESet:TYPE ...................................................... 174:SYSTem:PRESet:USER:SAVE ......................................... 174:SYSTem:REBoot ................................................................... 174:SYSTem:SHUTdown ............................................................ 174:SYSTem:TIME ........................................................................ 175:SYSTem:UPDate ................................................................... 175:SYSTem:VERSion:HARDware?........................................ 175:SYSTem:VERSion:SOFTware? ........................................ 175
S et:SYSTem:ALARm:STATe
Description

Sets the system alarm output on/off

Syntax:SYSTem:ALARm:STATe {OFF|ON|0|1}

Query Syntax:SYSTem:ALARm:STATe?

168

Query

エラー! スタイルが定義されていません。
Parameter

0
1
OFF
ON

Return parameter 0
1
ExampleTurn the alarm off.
Turn the alarm on.
Turn the alarm off.
Turn the alarm off.
The alarm is off.
The alarm is on.

:SYST:ALAR:STAT 1
S et
Query

:SYSTem:CLOCk<n>:DATE
Description

Sets the day for the selected wake-up clock.

Syntax:SYSTem:CLOCk<n>:DATE[MONday|TUESday|WEDnesday|THURsday|FRIday|
SATurday|SUNday]

Query Syntax:SYSTem:CLOCk<n>:DATE?

Parameter/<n>Return parameter MONday
TUESday
WEDnesday
THURsday
FRIdaySATurday
SUNdayExampleWake-up clock number 1 to 7
Set to Monday
Set to Tuesday
Set to Wednesday
Set to Thursday
Set to Friday
Set to Saturday
Set to Sunday:SYST:CLOC1:DATE MON
S et:SYSTem:CLOCk<n>:MODE

Query

Description

Sets the alarm mode for the selected wake-up clock.

Syntax:SYSTem:CLOCk<n>:MODE {REPeat|SINGle}

Query Syntax:SYSTem:CLOCk<n>:MODE?

169

GSP-930 User Manual<n>Parameter/
Return parameter
REPeat
SINGleExampleWake-up clock number 1 to 7
Set the wake-up clock to repeat.
Set the wake-up clock to single.

:SYST:CLOC1:MODE REP
S et
Query

:SYSTem:CLOCk<n>:STATe
Description

Turns the selected wake-up clock on/off.

Syntax:SYSTem:CLOCk<n>:STATe {OFF|ON|0|1}

Query Syntax:SYSTem:CLOCk<n>:STATe?

Parameter<n>Wake-up clock number 1 to 7

0
1
OFF
ON

Turn the wake-up clock off.
Turn the wake-up clock on.
Turn the wake-up clock off.
Turn the wake-up clock off.

Return parameter 0
1
ExampleThe wake-up clock is off.
The wake-up clock is on.

:SYST:CLOC1:STATE 1
S et
Query

:SYSTem:CLOCk<n>:TIME
Description

Sets the alarm time for the selected wake-up clock.

Syntax:SYSTem:CLOCk<n>:TIME <hour>,<minute>

Query Syntax:SYSTem:CLOCk<n>:TIME?

Parameter/<hour>
Return parameter <minute><n>
Example170<NR1> Sets the alarm hour.
<NR1> Sets the alarm minute.
Wake-up clock number 1 to 7

:SYST:CLOC1:TIME 20,50

エラー! スタイルが定義されていません。

:SYSTem:COMMunicate:GPIB[:SELF]:ADDRess

S et

Description

Sets the GPIB address.

Syntax:SYSTem:COMMunicate:GPIB[:SELF]:ADDRess<integer>

Parameter<integer>

Example:SYST:COMM:GPIB:ADDR 10

0 to 30:SYSTem:COMMunicate:LAN:ADDRessS et

Description

Sets the IP address.

Syntax:SYSTem:COMMunicate:LAN:ADDRess <ip address>

Parameter<ip address>

Example:SYST:COMM:LAN:ADDR 172.16.20.20

XXX.XXX.XXX.XXX:SYSTem:COMMunicate:LAN:MASKS et

Description

Sets the LAN mask.

Syntax:SYSTem:COMMunicate:LAN:MASK <ip address>

Parameter<ip address>

Example:SYST:COMM:LAN:MASK 172.16.20.20

XXX.XXX.XXX.XXX:SYSTem:COMMunicate:LAN:GATEway

S et

Description

Sets the LAN gateway.

Syntax:SYSTem:COMMunicate:LAN:GATEway <ip address>

Parameter<ip address>

Example:SYST:COMM:LAN:GATE 172.16.20.20

XXX.XXX.XXX.XXX

171

GSP-930 User Manual:SYSTem:COMMunicate:LAN:CONFigure

S etDescription

Configures the LAN to manual or DHCP mode.

Syntax:SYSTem:COMMunicate:LAN:CONFigure{DHCP|MANual}

Parameter/
DHCP
Return parameter MANual
ExampleConfigure the LAN to DHCP
Configure the LAN manually:SYST:COMM:LAN:CONF DHCPS et

:SYSTem:COMMunicate:LCI
Description

Configures the LAN to manual or DHCP mode.

Syntax:SYSTem:COMMunicate:LAN:CONFigure{DHCP|MANual}

Parameter/
DHCP
Return parameter MANual
ExampleConfigure the LAN to DHCP
Configure the LAN manually:SYST:COMM:LAN:CONF DHCP

:SYSTem:COMMunicate:SERial[:RECeive]
:BAUD

S et

Description

Sets the RS232 Baud rate.

Syntax:SYSTem: COMMunicate:SERial[:RECeive]:BAUD<integer>

Parameter<integer>

Example:SYST:COMM:SER:BAUD 9600

300|600|1200|2400|4800|9600|19200|
38400|57600|115200:SYSTem:COMMunicate:USB:MODE
Description
172

Configures the USB mode.

S etエラー! スタイルが定義されていません。
Syntax:SYSTem:COMMunicate:USB:MODE {HOST|DEVice}

Parameter/
HOST
Return parameter DEVice
ExampleUSB host mode
USB device mode

:SYST:COMM:USB:MODE DEV
S et
Query

:SYSTem:DATE
Description

Sets the system date.

Syntax:SYSTem:DATE <year>,<month>,<day>Query Syntax:SYSTem:DATE?

Parameter/<year>
Return parameter <month><day>
Example

<NR1>
<NR1>
<NR1>

:SYST:DATE 2011,03,27:SYSTem:ERRor:CLEar

S et

Description

Clears the errors messages from the error queue.

Syntax:SYSTem:ERRor:CLEar:SYSTem:ERRor[:NEXT]?

Query

Description

Returns the next message from the error queue.
Reading the error from the error queue will clear thaterror from the queue.

Syntax:SYST:ERR?

:SYSTem:KLOCk
Description

Locks/unlocks the front panel keys.

Syntax:SYSTem:KLOCk {ON|OFF}

S et

173

GSP-930 User Manual
Parameter

ON
OFF

Lock the front panel keys
Unlock the front panel keys

Example:SYST:KLOCK OFF
S et:SYSTem:PRESet
Description

Returns the GSP-930 to preset settings.

Syntax:SYST:PRES
S et
Query

:SYSTem:PRESet:TYPE
Description

Sets the preset type between user-defined andfactory default.

Syntax:SYSTem:PRESet:TYPE {USER|FACTory}

Query Syntax:SYSTem:PRESet:TYPE?

Parameter/
USER
Return parameter FACTory
ExampleUser defined preset
Factory default:SYST:PRES:TYPE USER:SYSTem:PRESet:USER:SAVES et

Description

Save the current environment as the “User” preset
settings.

Syntax:SYST:PRES:USER:SAVE:SYSTem:REBoot
Description

Restart/Reboot the GSP-930.

Syntax:SYSTem:REBoot:SYSTem:SHUTdown
174

S et

S et

エラー! スタイルが定義されていません。
Description

Shut down the GSP-930.

Syntax:SYST:SHUT
S et
Query

:SYSTem:TIME
Description

Sets the system time.

Syntax:SYSTem:TIME <hour>,<minute>,<second>

Query Syntax:SYSTem:TIME?

Parameter/<hour>
Return parameter <minute><second>
Example

<NR1>
<NR1>
<NR1>

:SYST:TIME 19,26,30
S et:SYSTem:UPDate
Description

Updates the system with new firmware from files
located on an external USB drive. The firmware files
must be included in the directory named /gsp930.

Syntax:SYST:UPD:SYSTem:VERSion:HARDware?
Description

Returns the system firmware version.

Query Syntax:SYSTem:VERSion:HARDware?

Return parameter <string>
ExampleT.1.X.X.X:SYST:VERS:HARD?
>T.1.0.0.0:SYSTem:VERSion:SOFTware?
Description

Query

Query

Returns the system software version.

175

GSP-930 User Manual
Query Syntax:SYSTem:VERSion:SOFTware?

Return parameter <string>
Example176

T1.00_2011.11.21_13:SYST:VERS:SOFT?
> T1.00_2011.11.21_13

エラー! スタイルが定義されていません。

STATus Commands:STATus:OPERation:CONDition? .....................................177:STATus:OPERation:ENABle .............................................170:STATus:OPERation[:EVENt]?..........................................170:STATus:OPERation:NTRansition ....................................170:STATus:OPERation:PTRansition ....................................171:STATus:QUEStionable:CONDition? ...............................171:STATus:QUEStionable:ENABle .......................................172:STATus:QUEStionable[:EVENt]? ....................................172:STATus:QUEStionable:NTRansition ..............................173:STATus:QUEStionable:PTRansition ..............................173:STATus:QUEStionable:FREQuency:CONDition?......174:STATus:QUEStionable:FREQuency:ENABle ..............174:STATus:QUEStionable:FREQuency[:EVENt]?...........174:STATus:QUEStionable:FREQuency:NTRansition .....175:STATus:QUEStionable:FREQuency:PTRansition .....175:STATus:QUEStionable:ACPLimit:CONDition?...........176:STATus:QUEStionable:ACPLimit:ENABle ...................176:STATus:QUEStionable:ACPLimit[:EVENt]?................177:STATus:QUEStionable:ACPLimit:NTRansition..........177:STATus:QUEStionable:ACPLimit:PTRansition..........178:STATus:PRESet ....................................................................178:STATus:OPERation:CONDition?

Query

Description

Returns the bit weight of the Operation Status
Condition register.

Query Syntax:STATus:OPERation:CONDition?

177

GSP-930 User Manual
Return parameter Bit
0~2
3
4
5
6~15
ExampleBit Weight
N/A
8
16
32
N/A

Description
Not used
Sweeping
Measuring
Wait for Trigger
Not used:STAT:OPER:COND?
>8
S et
Query

:STATus:OPERation:ENABle
Description

Sets or queries the Operation Status Event Enableregister.

Syntax:STATus:OPERation:ENABle <integer>

Query Syntax:STATus:OPERation:ENABle?

Return parameter Bit
0~2
3
4
5
6~15
ExampleBit Weight
N/A
8
16
32
N/A

Description
Not used
Sweeping
Measuring
Wait for Trigger
Not used:STAT:OPER:ENAB 32:STATus:OPERation[:EVENt]?

Query

Description

Returns the bit weight of the Operation Status Eventregister. Reading this register will clear the event
register.

Query Syntax:STATus:OPERation[:EVENt]?

178エラー! スタイルが定義されていません。
Return parameter Bit
0~2
3
4
5
6~15
ExampleBit Weight
N/A
8
16
32
N/A

Description
Not used
Sweeping
Measuring
Wait for Trigger
Not used:STAT:OPER?
>8
S et
Query

:STATus:OPERation:NTRansition
Description

Sets or queries the bit weight of the NTR filter for
the Operation Status register.

Syntax:STATus:OPERation:NTRansition <integer>

Query Syntax:STATus:OPERation:NTRansition?

Return parameter Bit
0~2
3
4
5
6~15
ExampleBit Weight
N/A
8
16
32
N/A

Description
Not used
Sweeping
Measuring
Wait for Trigger
Not used:STAT:OPER:NTR 32
S et:STATus:OPERation:PTRansition

Query

Description

Sets or queries the bit weight of the PTR filter for the
Operation Status register.

Syntax:STATus:OPERation:PTRansition <integer>

Query Syntax:STATus:OPERation:PTRansition?

179

GSP-930 User Manual
Return parameter Bit
0~2
3
4
5
6~15
ExampleBit Weight
N/A
8
16
32
N/A

Description
Not used
Sweeping
Measuring
Wait for Trigger
Not used:STAT:OPER:PTR 32
Query:STATus:QUEStionable:CONDition?
Description

Returns the bit weight of the Questionable Status
Condition register.

Query Syntax:STATus:QUEStionable:CONDition?

Return parameter Bit
5
8
9
10
11
12
13
ExampleBit Weight
16
128
512
1024
2048
4096
8192

Description
Frequency
Uncal
Limit Fail
ACPLimit
SEM Limit
TOI Limit
Pmet Limit Fail

:STAT:QUES:COND?
>16
S et

:STATus:QUEStionable:ENABle

Query

Description

Sets or queries the Questionable Status Event Enableregister.

Syntax:STATus:QUEStionable:ENABle <integer>

Query Syntax:STATus:QUEStionable:ENABle?

180

エラー! スタイルが定義されていません。
Return parameter Bit
5
8
9
10
11
12
13
ExampleBit Weight
16
128
512
1024
2048
4096
8192

Description
Frequency
Uncal
Limit Fail
ACPLimit
SEM Limit
TOI Limit
Pmet Limit Fail

:STAT:QUES:ENAB 4096
Query

:STATus:QUEStionable[:EVENt]?
Description

Returns the bit weight of the Questionable Status
Event register. Reading this register will clear the
event register.

Query Syntax:STATus:QUEStionable[:EVENt]?

Return parameter Bit
5
8
9
10
11
12
13
ExampleBit Weight
16
128
512
1024
2048
4096
8192

Description
Frequency
Uncal
Limit Fail
ACPLimit
SEM Limit
TOI Limit
Pmet Limit Fail

:STAT:QUES?
>16
S et

:STATus:QUEStionable:NTRansition

Query

Description

Sets or queries the bit weight of the NTR filter for
the Questionable Status register.

Syntax:STATus:QUEStionable:NTRansition <integer>

Query Syntax:STATus: QUEStionable:NTRansition?

181

GSP-930 User Manual
Return parameter Bit
5
8
9
10
11
12
13
ExampleBit Weight
16
128
512
1024
2048
4096
8192

Description
Frequency
Uncal
Limit Fail
ACPLimit
SEM Limit
TOI Limit
Pmet Limit Fail

:STAT:QUES:NTR 32
S et
Query

:STATus:QUEStionable:PTRansition
Description

Sets or queries the bit weight of the PTR filter for the
Questionable Status register.

Syntax:STATus:QUEStionable:PTRansition <integer>

Query Syntax:STATus: QUEStionable:PTRansition?

Return parameter Bit
5
8
9
10
11
12
13
ExampleBit Weight
16
128
512
1024
2048
4096
8192

Description
Frequency
Uncal
Limit Fail
ACPLimit
SEM Limit
TOI Limit
Pmet Limit Fail

:STAT:QUES:PTR 32:STATus:QUEStionable:FREQuency:
CONDition?

Query

Description

Returns the bit weight of the Questionable Status
Frequency Condition register.

Query Syntax:STATus:QUEStionable:FREQuency:CONDition?

182エラー! スタイルが定義されていません。
Return parameter Bit
5
ExampleBit Weight
16

Description
Invalid Span/BW:STAT:QUES:FREQ:COND?
>16
S et

:STATus:QUEStionable:FREQuency:ENABle

Query

Description

Sets or queries the Questionable Status Frequency
Event Enable register.

Syntax:STATus:QUEStionable:FREQuency:ENABle <integer>

Query Syntax:STATus:QUEStionable:FREQuency:ENABle?

Return parameter Bit
5
ExampleBit Weight
16

Description
Invalid Span/BW:STAT:QUES:FREQ:ENAB 16:STATus:QUEStionable:FREQuency[:EVENt]?

Query

Description

Returns the bit weight of the Questionable Status
Frequency Event register. Reading this register willclear the event register.

Query Syntax:STATus:QUEStionable:FREQuency[:EVENt]?

Return parameter Bit
5
ExampleBit Weight
16:STAT:QUES:FREQ?
>16:STATus:QUEStionable:FREQuency:
NTRansition
Description

Description
Invalid Span/BW

S et
Query

Sets or queries the bit weight of the NTR filter for
the Questionable Status Frequency register.
183

GSP-930 User Manual
Syntax:STATus:QUEStionable:FREQuency:NTRansition<integer>

Query Syntax:STATus:QUEStionable:FREQuency:NTRansition?

Return parameter Bit
5
ExampleBit Weight
16

Description
Invalid Span/BW:STAT:QUES:FREQ:NTR 16:STATus:QUEStionable:FREQuency:
PTRansition

S et
Query

Description

Sets or queries the bit weight of the PTR filter for the
Questionable Status Frequency register.

Syntax:STATus:QUEStionable:FREQuency:PTRansition<integer>

Query Syntax:STATus:QUEStionable:FREQuency:PTRansition?

Return parameter Bit
5
ExampleBit Weight
16

Description
Invalid Span/BW:STAT:QUES:FREQ:PTR 16:STATus:QUEStionable:ACPLimit:
CONDition?

Query

Description

Returns the bit weight of the Questionable Status
ACP Limit Condition register.

Query Syntax:STATus:QUEStionable:ACPLimit:CONDition?

184エラー! スタイルが定義されていません。
Return parameter Bit
0
1
2
3
4
5
6
7
ExampleBit Weight
1
2
4
8
16
32
64
128

Description
Main channel high fail
Main channel low failAdj1 high failAdj1 low failAdj2 high failAdj2 low failAdj3 high failAdj3 low fail

:STAT:QUES:ACPL:COND?
>1
S et:STATus:QUEStionable:ACPLimit:ENABle

Query

Description

Sets or queries the Questionable Status ACP Limit
Event Enable register.

Syntax:STATus:QUEStionable:ACPLimit:ENABle <integer>

Query Syntax:STATus:QUEStionable:ACPLimit:ENABle?

Return parameter Bit
0
1
2
3
4
5
6
7
ExampleBit Weight
1
2
4
8
16
32
64
128

Description
Main channel high fail
Main channel low failAdj1 high failAdj1 low failAdj2 high failAdj2 low failAdj3 high failAdj3 low fail

:STAT:QUES:ACPL:ENAB 3:STATus:QUEStionable:ACPLimit[:EVENt]?
Description

Query

Returns the bit weight of the Questionable Status
ACP limit Event register. Reading this register willclear the event register.
185

GSP-930 User Manual
Query Syntax:STATus:QUEStionable:ACPLimit[:EVENt]?

Return parameter Bit
0
1
2
3
4
5
6
7
ExampleBit Weight
1
2
4
8
16
32
64
128

Description
Main channel high fail
Main channel low failAdj1 high failAdj1 low failAdj2 high failAdj2 low failAdj3 high failAdj3 low fail

:STAT:ACPL:QUES?
>3:STATus:QUEStionable:ACPLimit:
NTRansition

S et
Query

Description

Sets or queries the bit weight of the NTR filter for
the Questionable Status ACP Limit register.

Syntax:STATus:QUEStionable:ACPLimit:NTRansition<integer>

Query Syntax:STATus:QUEStionable:ACPLimit:NTRansition?

Return parameter Bit
0
1
2
3
4
5
6
7
Example186

Bit Weight
1
2
4
8
16
32
64
128

:STAT:QUES:ACPL:NTR 3

Description
Main channel high fail
Main channel low failAdj1 high failAdj1 low failAdj2 high failAdj2 low failAdj3 high failAdj3 low fail

エラー! スタイルが定義されていません。

:STATus:QUEStionable:ACPLimit:
PTRansition

S et
Query

Description

Sets or queries the bit weight of the PTR filter for the
Questionable Status ACP Limit register.

Syntax:STATus:QUEStionable:ACPLimit:PTRansition<integer>

Query Syntax:STATus:QUEStionable:ACPLimit:PTRansition?

Return parameter Bit
0
1
2
3
4
5
6
7
ExampleBit Weight
1
2
4
8
16
32
64
128

Description
Main channel high fail
Main channel low failAdj1 high failAdj1 low failAdj2 high failAdj2 low failAdj3 high failAdj3 low fail

:STAT:QUES:ACPL:PTR 3:STATus:PRESet
Description

Loads the preset settings.

Syntax:STATus:PRESet

S et

TRACe Commands:TRACe[:DATA]? ....................................................................179:TRACe[:DATA]?
Description

Query

Returns the trace data for the selected trace in CSV
format.
187

GSP-930 User Manual
Query Syntax:TRACe[:DATA]? <trace name>Parameter<trace name> [trace1|trace2|trace3|trace4|trace5]

Return Parameter <data>
Example188

Trace data in CSV format:TRACe[:DATA]? trace1>-5.234e+01,-4.593e+01,-5.533e+01,-4.604e+01,>5.353e+01,-4.557e+01,-5.280e+0>1,-4.785e+01,-5.459e+01,-4.578e+01,………………

エラー! スタイルが定義されていません。

TRIGger Commands:TRIGger[:SEQuence]:DELay.............................................180:TRIGger[:SEQuence]:DEMod:DELay..............................180:TRIGger[:SEQuence]:DEMod:LEVel...............................181:TRIGger[:SEQuence]:DEMod:MODE..............................181:TRIGger[:SEQuence]:DEMod:SLOPe ............................181:TRIGger[:SEQuence]:DEMod:SOURce .........................182:TRIGger[:SEQuence]:EXTernal:SLOPe ........................182:TRIGger[:SEQuence]:MODE .............................................182:TRIGger[:SEQuence]:PMETer:SOURce .......................183:TRIGger[:SEQuence]:SOURce.........................................183:TRIGger[:SEQuence]:VIDeo:FREQuency.....................183:TRIGger[:SEQuence]:VIDeo:LEVel.................................184:TRIGger[:SEQuence]:VIDeo:SLOPe ..............................184
S et
Query

:TRIGger[:SEQuence]:DELay
Description

Sets the trigger delay time in seconds.

Syntax:TRIGger[:SEQuence]:DELay <time>

Query Syntax:TRIGger[:SEQuence]:DELay?

Parameter/<time>
Return parameter
ExampleDelay time in seconds:TRIG:DEL 1.0e-2
S et:TRIGger[:SEQuence]:DEMod:DELay

Query

Description

Sets the AF trigger delay time in seconds for AM/FM
demodulation.

Syntax:TRIGger[:SEQuence]:DEMod:DELay <time>

Query Syntax:TRIGger[:SEQuence]:DEMod:DELay?

189

GSP-930 User Manual
Parameter/<time>
Return parameter
ExampleDelay time in seconds:TRIG:DEM:DEL 1.0 ms
S et
Query

:TRIGger[:SEQuence]:DEMod:LEVel
Description

Sets the trigger level for AM/FM demodulation.

Syntax:TRIGger[:SEQuence]:DEMod:LEVel <integer>

Query Syntax:TRIGger[:SEQuence]:DEMod:LEVel?

Parameter/<integer>
Return parameter
ExampleTrigger level in %:TRIG:DEM:LEV 10
S et
Query

:TRIGger[:SEQuence]:DEMod:MODE
Description

Sets the triggering mode for the AF Trigger in AM/FM
demodulation.

Syntax:TRIGger[:SEQuence]:DEMod:MODE{NORMal|SINGle|CONTinuous}

Query Syntax:TRIGger[:SEQuence]:DEMod:MODE?

Parameter/
NORMalReturn parameter SINGle
CONTinuous
ExampleNormal trigger mode
Single trigger
Continuous trigger:TRIG:DEM:MODE CONT
S et

:TRIGger[:SEQuence]:DEMod:SLOPe
Description

Sets the trigger slope.

Syntax:TRIGger[:SEQuence]:DEMod:SLOPe{POSitive|NEGative}

Query Syntax:TRIGger[:SEQuence]:DEMod:SLOPe?

190

Query

エラー! スタイルが定義されていません。
Parameter/
POSitive
Return parameter NEGative
ExamplePositive slope
Negative slope:TRIG:DEM:SLOP POS:TRIGger[:SEQuence]:DEMod:SOURce

S etDescription

Sets the triggering source for AM/FM demodulation

Syntax:TRIGger[:SEQuence]:DEMod:SOURce{IMMediate|VIDeo}

Parameter

IMMediate
VIDeo

Example:TRIG:DEM:SOUR IMM

Free run trigger
Trigger on the video signal level

S et
Query

:TRIGger[:SEQuence]:EXTernal:SLOPe
Description

Sets the external trigger slope

Syntax:TRIGger[:SEQuence]:EXTernal:SLOPe{POSitive|NEGative}

Query Syntax:TRIGger[:SEQuence]:EXTernal:SLOPe?

Parameter/
POSitive
Return parameter NEGative
ExamplePositive slope
Negative slope:TRIG:EXT:SLOP POS
S et:TRIGger[:SEQuence]:MODE
Description

Sets the triggering mode.

Syntax:TRIGger[:SEQuence]:MODE{NORMal|SINGle|CONTinuous}

Query Syntax:TRIGger[:SEQuence]:MODE?

Query

191

GSP-930 User Manual
Parameter/
NORMalReturn parameter SINGle
CONTinuous
ExampleNormal trigger mode
Single trigger
Continuous trigger:TRIG: MODE CONT
S et
Query

:TRIGger[:SEQuence]:PMETer:SOURce
Description

Sets the triggering source to immediate or external

Syntax:TRIGger[:SEQuence]:PMETer:SOURce{IMMediate|EXTernal}

Query Syntax:TRIGger[:SEQuence]:PMETer:SOURce?

Parameter/
IMMediate
Return parameter EXTernal
ExampleFree run trigger
External trigger:TRIG:PMET:SOUR IMM
S et:TRIGger[:SEQuence]:SOURce

Query

Description

Sets the triggering source to immediate, external orvideo.

Syntax:TRIGger[:SEQuence]:SOURce{IMMediate|EXTernal|VIDeo}

Query Syntax:TRIGger[:SEQuence]:SOURce?

Parameter/
IMMediate
Return parameter EXTernal
VIDeo
ExampleFree run trigger
External trigger
Video trigger:TRIG:SOUR IMM
S et:TRIGger[:SEQuence]:VIDeo:FREQuency

Query

Description

Sets the video trigger frequency.

Syntax:TRIGger[:SEQuence]:VIDeo:FREQuency <freq>192エラー! スタイルが定義されていません。
Query Syntax:TRIGger[:SEQuence]:VIDeo:FREQuency?

Parameter/<freq>
Return parameter
Example<NR3> frequency in Hz.

:TRIG:VID:FREQ?
>2.5e+6
S et
Query:TRIGger[:SEQuence]:VIDeo:LEVel
Description

Sets the video trigger level.

Syntax:TRIGger[:SEQuence]:VIDeo:LEVel <ampl>

Query Syntax:TRIGger[:SEQuence]:VIDeo:LEVel?

Parameter/<ampl>
Return parameter
Example<NR3> amplitude in dBm.

:TRIG:VID:LEV 10
S et:TRIGger[:SEQuence]:VIDeo:SLOPe
Description

Sets the video trigger slope

Syntax:TRIGger[:SEQuence]:VIDeo:SLOPe{POSitive|NEGative}

Query Syntax:TRIGger[:SEQuence]:VIDeo:SLOPe?

Parameter/
POSitive
Return parameter NEGative
ExampleQuery

Positive slope
Negative slope:TRIG:VID:SLOP POS

193

GSP-930 User Manual

UNIT Commands:UNIT:PMETer:POWer........................................................... 185:UNIT:POWer............................................................................ 185
S et
Query

:UNIT:PMETer:POWer
Description

Sets the amplitude unit used for the Power Meter
mode.

Syntax:UNIT:PMETer:POWer {DBM|MW}

Query Syntax:UNIT:PMETer:POWer?

Parameter/
DBM
Return parameter MW
ExampleDecibelsMilliwatts:UNIT:PMET:POW DBM
S et
Query

:UNIT:POWer
Description

Sets the amplitude unit used for the Spectrum mode.

Syntax:UNIT:POWer {DBM|DBMV|DBUV|W|V}

Query Syntax:UNIT:POWer?

DBM
Parameter/
Return parameter DBMV
DBUVWVExample194

Decibels
decibels relative to one millivolt
decibels relative to one microvolt
Watt
Volt

:UNIT:POW DBM