3D Accuitomo
170Thinking ahead. Focused on life.

3D Accuitomo

80 μm for unsurpassed image clarity
The 3D Accuitomo offers unsurpassed high resolution images with wide fields of view.
Its super-fine minimal voxel size of just 80 μm allows diagnosing even the most subtle details of the temporal bone, nasal
cavities, paranasal sinuses, mandible, and teeth.

Ø40 × H 40 mm (80 μm)

Unsurpassed high resolution image
with minimal voxel size of 80 μm
The minimum voxel size of 80 μm
ensures clear, high resolution images
even when magnified.

Ø170 × H 120 mm (250 μm)

Ø170 × H 120 mm (250 μm)

High Resolution Mode (80 μm)

3D Accuitomo 17023

Various Fields of View

9 fields of view for flexible scanning from local to large areas

High resolution even at large FOVs

The 3D Accuitomo is equipped with 9 FOVs (fields of view) that allows flexibility when scanning patients with a variety ofdiagnostic needs and clinical indications, from a large area (Ø170 × H 230 mm) that covers the maxillofacial region to a local
area (Ø40 × H 40 mm).
Reducing exposure dose is possible by selecting the most suitable FOV.

The minimum voxel size can be selected from 80 μm, 100 μm, 125 μm, 200 μm, or 250 μm depending on your diagnostic
needs and clinical indications.
The 3D Accuitomo is able to provide high resolution with less distortion, even at large FOVs.
FOV can be offset so that even the temporal bone region can be positioned at the center of the FOV. This results in wellfocused, high resolution images.

Ø100 × H 100 mm
Ø170 × H 120 mm

Ø170 mm

Ø140 × H 100 mm

Ø140 mm

Ø170 × H 50 mm

Ø100 mm

Ø80 × H 80 mm
Ø100 × H 50 mm

Ø80 mm
Ø60 mmØ60 × H 60 mm

Ø40 mmØ140 × H 50 mm
Ø40 × H 40 mm

Ø140 × H 100 mm（(200 μm)）

Ø100 × H 100 mm (160 μm)

Ø60 × H 60 mm (100 μm)

Ø40 mmZoom reconstruction from original data
The 3D Accuitomo is equipped with a unique zoom reconstruction function allowing you to zoom in and reconstruct a newvolume from the original scan, without the need for additional acquisitions. The new volume can be reconstructed with aresolution of up to 80μm improving diagnostic accuracy with no additional X-ray exposure to the patient.
Fields of View
FOV406080
100140Voxel Size

Ø40 × H 40 mm

80 μm

Ø60 × H 60 mm

100 μmØ80 × H 80 mm

125 μmØ100 × H 50 mm
Ø100 × H 100 mm
Ø140 × H 50 mm
Ø140 × H 100 mm
Ø170 × H 50 mm
Ø170 × H 120 mm

406080100140160 μm

200 μm

250 μmStandard Mode Ø170 mm × H 120 mm
Ø170 × H 120 mm (250 μm)
4Ø40 × H 40 mm( 80 μm)
5Adaptable Acquisition Modes

Four exposure modes, from high resolution to high speed

Fulfilling supportive functions for clinical practices

High Resolution(Hi-Res) Mode : Pixel size of the flat-panel detector is 1/4 compared to standard mode. This mode has the best spatialresolution.
High Fidelity(Hi-Fi) Mode: Higher data density for clearer image than standard mode. This mode is suitable for the zoom reconstruction function.
Standard(Std) Mode
: Suitable for all applications; from local to large area such as temporal bone, nasal cavity, jawbone, teeth.
High Speed (Hi-Speed) Mode : This helps reduce motion artifacts during the scan. Suitable for patients such as children who have difficulty
controlling movements.

The 3D Accuitomo is equipped with an application that has various functions that allow simulations within volume rendering.

Mastoidectomy Mode (neural tubes drawing and CT volume removing)Standard Mode Ø170 mm × H 120 mm

High Fidelity Mode Ø80 mm × H 80 mm

360° Full Scan

180° Half Scan

High Resolution (Hi-Res) Mode

30.8 sec.

15.8 sec.

High Fidelity (Hi-Fi) Mode*

30.8 sec.

15.8 sec.

Standard (Std) Mode

17.5 sec.

9.0 sec.

High Speed (Hi-Speed) Mode

10.5 sec.

5.4 sec.

*Pseudo Rigid Scope Mode (perspective projection)

360° Full Scan

30.8 sec.

High Resolution Mode & High Fidelity Mode360°

17.5 sec.

Standard Mode

10.5 sec.

High Speed Mode
180° Half Scan

180°

9.0 sec.

Standard ModeHigh Speed Mode

5.4 sec.

*High resolution mode and high speed mode are only available for Ø40 × H 40 mm and Ø60 × H 60 mm FOVs.

67

Case Example 1

Case Example 2

Case: Implantology
Female patient referred for 3-dimensional
analysis of esthetic complications after implant
treatment in the left maxillary incisor region(Photo 1A).
The clinical status exhibits a mucosal recession
as well as a flattening and discoloration of thefacial mucosa at the implant crown. The patientcomplained about recurrence of the periimplant infections.

Case: Otosclerosis
The otospongiotic lesion is detected around theanterior oval window (fissula ante fenestram).
MPR images created in the planes parallel to thestapes superstructure clearly show the positional
relationship between the stapes and theotosclerotic lesion.

Symbols “*” in the photos 1C and 1D indicatesthe location of nasopalatine duct.
Image Courtesy: Prof. em. Dr. Daniel BuserProf. Dr. Michael Bornstein

Photo 1A:
Clinical aspect at the initial examination

Image Courtesy: Kawano Ear Surge Clinic

Photo 1C:
Axial image

Case: Odontogenic maxillary sinusitis
Here is a high-resolution CBCT scan of the left firstmolar of the maxilla which has undergone
endodontics (root canal treatment and crownrestoration). The floor of the left maxillary antrum
shows the early stages of odontogenic maxillarysinusitis due to an apical lesion of the left firstmolar.
Image Courtesy: Sato Clinic
8

Photo 1B:
Coronal image

Photo 1D:
Sagittal image
9

Specifications

Trade Name

3D Accuitomo(Unit: mm)

2000

Type

EX1/2 F17Power Supply

AC 100/110/120 V

1250

MCT-11800

Model300

XYZ Slice View TomographAC 220/230/240 VAC
Power Consumption

Max. 2.0 kVA
1620

Dimensions
Main Unit

850540W 1,620 mm × D 1,250 mm × H 2,080 mm(63-3/4" × 49-1/4" × 82")
W 100 mm × D 40 mm × H 115 mm
2080

Control Box

Weight:Approx. 400 kg (Approx. 882 lbs)

1500

(4" × 1-5/8" × 4-1/2")

X-ray HeadHigh Resolution (Hi-Res) ModeHigh Speed (Hi-Speed) Mode

Equipment

Exposure ModeHigh Fidelity (Hi-Fi) Mode

Zoom Reconstruction
Two Direction Scout
Scan Mode

3D ViewerØ80 × H 80 mm (125 μm)
Ø100 × H 100 mm (160 μm)
Ø140 × H 50 mm (200 μm)

SoftwareEquipment

Ø60 × H 60 mm (125 μm)

Fields of View
(Voxel Size)

Tube Current

1 – 10 mA(Max. 8 mA: Hi-Fi, Hi-Res Mode)360°

Focal Spot Size

0.5

Exposure Time

Std Mode: 17.5/9.0 sec.
Hi-Fi Mode: 30.8/15.8 sec.

180°

Hi-Res Mode: 30.8/15.8 sec.

Volume rendering

Ø40 × H 40 mm (80 μm)

Ø100 × H 50 mm (160 μm)

60 – 90 kVSpecifications

Specifications
Standard (Std) Mode

Tube Voltage

Data ExportØ140 × H 100mm (200 μm)
Ø170 × H 50 mm (250 μm)

DICOM Storage

Ø170 × H 120 mm (250 μm)

Print Center

Hi-Speed Mode: 10.5/5.4 sec.

CrvdMPRImage Carving

Field of View

Ø40 × H 40 mm, Ø60 × H 60 mm,

Neural Tube Drawing

Ø80 × H 80 mm,

One Data Viewer

Ø100 × H 50 mm, Ø100 × H 100 mm,

One Volume Viewer

Ø140 × H 50 mm, Ø140 × H 100 mm,

DICOM File ExportØ170 × H 50 mm, Ø170 × H 120 mm,
Voxel Size

80 μm/125 μm/160 μm/250 μm* X-ray protection should be provided for the patient when X-rays are emitted.

1011Diagnostic and Imaging Equipment
Treatment Units
Handpieces and Instruments
Endodontic System
Laser Equipment
Laboratory Devices
Development and Manufacturing
J. MORITA MFG. CORP.
680 Higashihama Minami-cho, Fushimi-ku,
Kyoto 612-8533, Japan
T +81. (0)75. 611 2141, F +81. (0)75. 622 4595
Morita Global Website
www.morita.com
Distribution
J. MORITA USA, INC.
9 Mason, lrvine CA 92618, USA
T +1. 949. 581 9600, F +1. 949. 581 8811
J. MORITA EUROPE GMBH
Justus-von-Liebig-Strasse 27b, 63128 Dietzenbach, Germany
T +49. (0)6074. 836 0, F +49. (0)6074. 836 299
MORITA DENTAL ASIA PTE. LTD.
150 Kampong Ampat#06-01A KA Centre, Singapore 368324
T +65. 6779. 4795, F +65. 6777. 2279
J. MORITA CORP. AUSTRALIA & NEW ZEALAND
Suite 2.05, 247 Coward Street, Mascot NSW 2020, Australia
T +61. (0)2. 9667 3555, F +61. (0)2. 9667 3577
J. MORITA CORP. MIDDLE EAST
4 Tag Al Roasaa, Apartment 902, Saba Pacha 21311 Alexandria, Egypt
T +20. (0)3. 58 222 94, F +20. (0)3. 58 222 96
J. MORITA CORP. INDIA
Filix Office No.908, L.B.S. Marg, Opp. Asian Paints, Bhandup (West), Mumbai 400078, India
T +91-22-2595-3482
J. MORITA MFG. CORP. INDONESIA
28F, DBS Bank Tower, Jl. Prof. Dr. Satrio Kav. 3-5, Jakarta 12940, Indonesia
T +62-21-2988-8332, F + 62-21-2988-8201
SIAMDENT CO., LTD.
71/10 Mu 5, Thakham, Bangpakong, Chachuengsao 24130, Thailand
T +66. 38. 573042, F +66. 38. 573043
www.siamdent.com

Subject to technical changes and errors.

Pub. No.: MEDP.CT.05.01

JME EN 0921

Educational and Training Systems
Auxiliaries