Pneumatic Instruments Catalog

Invensys plc
3rd Floor
40 Grosvenor Place
LondonSW1X 7AW
United Kingdom
Tel: +44 (0)203 155 1200
www.invensys.com

Foxboro33 Commercial Street
Foxboro, Massachusetts 02035-2099 U.S.A.
+1(866) 746-6477 Toll-free within U.S.A+1(508) 549-2424 Global

Foxboro Eckardt
Pragstrasse 82
D-70376 Stuttgart
GermanyTel +49 (0)711 502-0
Fax +49 (0)711 502-597
Email: salessupport@foxboro-eckardt.deInvensys, the Invensys logo, ArchestrA, Avantis, Eurotherm, Foxboro, IMServ, InFusion, SimSci-Esscor, Skelta, Triconex, and Wonderware are trademarks of Invensys plc, its subsidiaries oraffiliates. All other brands and product names may be the trademarks or service marks of their representative owners.
© 2013 Invensys Systems, Inc. All rights reserved. No part of the material protected by this copyright may be reproduced or utilized in any form or by any means, electronic or mechanical,
including photocopying, recording, broadcasting, or by any information storage and retrieval system, without permission in writing from Invensys Systems, Inc.
PN FX-0180

Rev. 04/13

OrderingSpecifying Your Instrument or Accessory
You can easily specify many instruments and accessories described in thiscatalog. Sections covering our most popular items include all the technical data
you need to know for most applications. To specify the appropriate item, simply
follow the step-by-step procedure at the end of each description. Your Foxboro
representative can help also.
For Example:
Choose the product code.
Itemize the information needed to complete the specification.
Ordering:
To order, contact your Foxboro representative.
Or call, in North America: 1-866-746-6477, or International: 001-508-549-2424.
Or visit www.buyautomation.com/officelocator.
Give us the ordering information using this catalog. Then tell us where you want
the order shipped and when you need it delivered. We’ll do the rest.
Need Help?
If you have questions, let us know. We’re ready, and eager, to assist you.
Applications are a Foxboro specialty. We can help you figure out the best
instrument for your purpose.
Instrument catalog available — ask your Foxboro representative about it.

1) SPECIFY MODEL NUMBER: 13
Transmitter:

–

–

–

316 ss body.......................................................................................................... A
Monel body.......................................................................................................... A4
Hastelloy C body (supplied without process connectors)............................. A5

Span Limits:

5 and 62 kPa or 20 and 250 inH2O ΔP....................................................................... M
50 and 210 kPa or 200 and 850 inH2O ΔP................................................................ H

Body Material:

316 ss (13A only)..................................................................................................................... S
Monel (13A4 only)................................................................................................................... M
Hastelloy C (13A5 only) (Process Connectors code 0 below must be specified........... C

Process Connectors:

1/4 NPT..............................................................................................................................................
1/2 NPT..............................................................................................................................................
R1/4....................................................................................................................................................
R1/2....................................................................................................................................................
Welding neck for 14 x 21 mm tube and 1/2 in Schedule 80 pipe............................................
None, body tapped for 1/4 NPT....................................................................................................

1
2
3
4
6
0

Optional Features: (also see previous page)

Zero elevation kit........................................................................................................................................ L
Zero suppression kit.................................................................................................................................. R

2) SPECIFY OUTPUT SIGNAL___________________________________________________
3) SPECIFY CALIBRATED DIFFERENTIAL PRESSURE RANGE_______________________
4) SPECIFY ADDITIONAL OPTIONAL FEATURES_________________________________
5) SPECIFY INFORMATION FOR INSTRUMENT TAG______________________________

33 Commercial Street
Foxboro, Massachusetts 02035-2099 U.S.A.

Your Foxboro representative:

Table of Contents

Table of Contents

1

Pneumatic Transmitters..................................................................Page 2
Differential Pressure............................................................................................... 3
Flow........................................................................................................................... 9
Pressure.................................................................................................................. 14
Repeater................................................................................................................. 18
Liquid Level and Density...................................................................................... 20
Temperature.......................................................................................................... 28
Filter Regulators..................................................................................................... 31
Miscellaneous........................................................................................................ 32

Primary Measurement Elements............................................................ 33
Orifice Plates and Flange Unions........................................................................ 29
IFOA Integral Flow Orifice Assemblies.............................................................. 36
Diaphragm Pressure Seals................................................................................... 39
Thermowells and DEWCEL Dew Point Measurement Systems...................... 40

Large Case Instruments.............................................................................. 41

3

Controllers.............................................................................................................. 42
Recorders and Indicators..................................................................................... 51
Recorders and Controllers................................................................................... 54
Elements................................................................................................................. 56
Miscellaneous........................................................................................................ 65

Panel-Mounted Pneumatic Instruments............................................. 66

4

Indicating Controllers........................................................................................... 67
Recorders and Controllers................................................................................... 73
Loading, Switching and Bias................................................................................ 76
Subpanel................................................................................................................. 78
Computing Elements............................................................................................ 79
Computing Relay................................................................................................... 81
Miscellaneous........................................................................................................ 83

Converters......................................................................................................... 84

5

Current-to-Pneumatic........................................................................................... 85
Control Valves – Actuators................................................................................... 90
Control Valves – Positioners................................................................................. 91

Accessories and Supplies........................................................................... 92

6

Humitex Circular and Strip Charts....................................................................... 92
Pens and Pen Arms............................................................................................... 94

Product Index.............................................................................................99

Page 1

Pneumatic Transmitters

Foxboro Pneumatic Transmitters are the industry standard.
All of these instruments have been field proven through years of experience.
They are precise, easy to maintain, and available in a wide range of materials.
The Foxboro line encompasses a variety of measurement types.

Flow Measurement
Differential pressure transmitters.

Temperature Measurement
Filled thermal systems and thermocouple/ RTD devices.

Pressure Measurement
Force balance and indicating transmitters.

Liquid Level Measurement
Differential pressure transmitters with or without chemical seals and buoyancytransmitters.

Other Applications
Sanitary designs, relative humidity trans­mitters, speed transmitters, and repeaters.
Foxboro is the leading single source of pneumatic transmitters for the processindustries.
No other manufacturer makes as many types for as many applications.

Page 2

Pneumatic Transmitters (Differential Pressure)
15A, 13A, 13H Series
Pneumatic d/p Cell
Transmitters

15A, 13A, 13H

n 15A—low range transmitter adjustable forranges from 0-1.3 to 0-6 kPa (0-5 to 0-25
inH20) ΔP at static pressures up to 3.5
MPa (500 psi)

n 13A—adjustable for ranges from 0-5 to

0-210 kPa (0-20 to 0-850 inH20) ΔP at
static pressures up to 14 MPa (2000 psi)

13A

n 13H—adjustable for ranges from 0-5 to

0-210 kPa (0-20 to 0-850 inH20) ΔP at
static pressures up to 40 MPa (6000 psi)n Low Air Consumption

For complete specifications, refer to
Product Specification Sheet PSS 2B-1C1 A.

13H

15A

The 15A, 13A, 13H Series Pneumatic d/p
Cell Transmitters measure differential
pressure and transmit a proportional
pneumatic output signal.

Functional Specifications
Static Pressure, Span, and Range Limits
Static Pressure Limit
Series

MPa

psi

15A
13A
13H

3.5
14
40

500
2000
6000
Span Limits		
Series

Span
Code

kPa
ΔP

inH2O
ΔP

15A

L

1.3 and 6

5 and 25

13A and
13H

M
H

5 and 62
50 and 210

20 and 250
200 and 850

Process Temperature Limits:-40 and +120°C (-40 and +250°F) at capsule.
Ambient Temperature Limits:-40 and +120°C (-40 and +250°F).

Range Limits (a)		Series

Span
Code

kPa
ΔP

inH2O
ΔP

15A

L

±12.5

±50

13A and
13H

M
H

±62
±210

±250
±850

(a) Nonzero-based ranges require optional zero elevationor suppression kit. See “HOW TO ORDER.” Upper andlower range values must not exceed range limits. Negative
numbers indicate a higher pressure on the normal “LowSide” of the transmitter, such as may occur in closed tankapplication with a wet leg.

Output Signal and Suppiy Pressure:Output Signal

Supply Pressure

20 to 100 kPa
3 to 15 psi

140 kPa
20 psi

Performance Specifications
Accuracy:

15A Series: ±0.5% of calibrated span.
13A and 13H Series: ±0.5% of calibrated span
except +0.75% of calibrated span for calibrated
spans greater than 130 kPa or 525 inH20 ΔP.

Repeatability:

0.1% of calibrated span.
Hysteresis:

0.1% of calibrated span.

Page 3

Pneumatic Transmitters (Differential Pressure)
15A, 13A, 13H

Physical Specifications
Materials of Wetted Parts:
Series		
Wetted Part

15A, 13A, 13HA

13A4

13A5

Body
Capsule Diaphragm
Other Capsule Parts
Force Bar
Force Bar Seal
Force Bar Gasket
Capsule Gaskets
Process Connection Gaskets

AISI 316 ss
316L ss
316 ss
316 sscobalt-nickel
silicone elastomer (b)316 ssptfe (c)Monel
Monel 400 (a)
Monel
Monel
cobalt-nickel
Viton-A
Monel
ptfe

Hastelloy C
Hastelloy C276
Hastelloy C
Hastelloy C
cobalt-nickel
Viton-Aptfe(d)(a) Duranickel capsule diaphragm material supplied when Span Limits code H is specified.
(b) 13H Series uses Buna-N as standard.
(c) 13H Series uses glass filled ptfe as standard.
(d) Hastelloy C bodies are supplied without process connectors.

Mounting:
Direct to process or by bracket for DN 50 or 2 inhorizontal or vertical pipe. Bracket is always
supplied.
Enclosure Classification:
Meets IEC IP53 and provides the environmental
protection of NEMA Type 3.

Materials:
Body Bolts: Alloy steel per ASTM A193 GradeB7; or 17-4 with Hastelloy C bodies.
Cover:High impact, glass filled polycarbonate.
Air Connections:
The supply and output connections are tappedfor 1/4 NPT.

Optional Features
Capsule Materials:
Hastelloy C, Monel, Duranickel, and tantalum.

Air Supply Set:
Adjustable regulators with or without gauge.

HF Alkylation Service:
Maximum life with economical materials.

High Process Temperature:
Body temperature up to 190°C (375°F).

Chlorine Service Preparation:
Cleaned for chlorine service.

Integral Orifice:For low flow measurements. Refer to Page 31.

Hydrogen Service Preparation:
Gold-plated capsule to prevent hydrogen
penetration.

Bypass Manifolds:
Refer to Page 27.

High Damping:
Medium and low range capsules only.
NACE:
Compliance to NACE Standard MR-01-75.

Page 4Oxygen Service Preparation:
Cleaned and packaged for oxygen service.
Lower Spans:
Provides spans as low as one-half standard.

Pneumatic Transmitters (Differential Pressure)
15A, 13A, 13H

How to Order
1) SPECIFY MODEL NUMBER: 15		

- L			 -		Transmitter:

316 ss body....................................................................................................................A

Span Limits:

1.3 and 6 kPa or 5 and 25 inH20 ΔP......................................................................................L

Body Material:

316 ss (15A only)............................................................................................................................. S

Process Connectors:

1/4 NPT......................................................................................................................................................1
1/2 NPT......................................................................................................................................................2
R1/4 (metric).............................................................................................................................................3
R1/2 (metric).............................................................................................................................................4
Welding neck for 14 x 21 mm tube and 1/2 in Schedule 80 pipe....................................................6
None, body tapped for 1/4 NPT............................................................................................................0

Optional Features: (also see previous page)

Zero elevation kit.............................................................................................................................................. L
Zero suppression kit........................................................................................................................................ R

1) SPECIFY MODEL NUMBER: 13____ - ___ ___ ___ - ___Transmitter:

316 ss body....................................................................................................................A
Monel body...................................................................................................................A4
Hastelloy C body (supplied without process connectors)......................................A5

Span Limits:

5 and 62 kPa or 20 and 250 inH20 ΔP.................................................................................M
50 and 210 kPa or 200 and 850 inH20 ΔP........................................................................... H

Body Material:

316 ss (13A only)............................................................................................................................. S
Monel (13A4 only).......................................................................................................................... M
Hastelloy C (13A5 only) (Process Connectors code 0 below must be specified)..................C

Process Connectors:

1/4 NP........................................................................................................................................................1
1/2 NP........................................................................................................................................................2
R1/4............................................................................................................................................................3
R1/2............................................................................................................................................................4
Welding neck for 14 x 21 mm tube and 1/2 in Schedule 80 pipe....................................................6
None, body tapped for 1/4 NPT............................................................................................................0

Optional Features: (also see previous page)

Zero elevation kit...............................................................................................................................................L
Zero suppression kit.........................................................................................................................................R

HOW TO ORDER Continued on Next Page.

Page 5

Pneumatic Transmitters (Differential Pressure)
15A, 13A, 13H

How to Order (continued)
1) SPECIFY MODEL NUMBER: 13H
Transmitter:____ - ___ ___ ___ - ___316 ss body .................................................................................................................. A

Span Limits:

5 and 62 kPa or 20 and 250 inH20 ΔP.................................................................................M
50 and 210 kPa or 200 and 850 inH20 ΔP........................................................................... H

Body Material:

316 ss (13HA only).......................................................................................................................... S

Process Connectors: (a)1/4 NPT......................................................................................................................................................1
1/2 NPT......................................................................................................................................................2
R1/4 ........................................................................................................................................................3
R1/2 ........................................................................................................................................................4
None, body tapped for 9/16-18 Aminco fitting...................................................................................5
None, body tapped for 1/4 NPT............................................................................................................0

Optional Features: (also see Page 4)

Zero elevation kit.............................................................................................................................................. L
Zero suppression kit........................................................................................................................................ R(a) The transmitter is normally supplied with body tapped for 1/4 NPT unless Process Connectors code
5 is specified.

2) SPECIFY OUTPUT SIGNAL__________________________________________________________
3) SPECIFY CALIBRATED DIFFERENTIAL PRESSURE RANGE_____________________________
4) SPECIFY ADDITIONAL OPTIONAL FEATURES________________________________________
5) SPECIFY INFORMATION FOR INSTRUMENT TAG____________________________________

Page 6

Pneumatic Transmitters (Differential Pressure)
11DM Series Pneumatic
Differential Pressure
Transmitters

11DM

n Differential Pressure Measurement —ranges adjustable from 0-0.14 to 0-14.0
MPa (0-20 to 0-2000 psi) ΔP

n Low Air Consumption

For complete specifications, refer to
Product Specification Sheet PSS 2B-1A2 A.

The 11 DM Series Pneumatic Differential
Pressure Transmitters measure differential
pressure and transmit a proportional
pneumatic signal.

Functional Specifications
Span and Range Limits:
Span Limits
Span Limits CodeMPa ΔP

psi ΔP

B
C
D
E

0.14 and 1.4
0.3 and 2.8
0.7 and 7.0
1.4 and 14.0

20 and 200
40 and 400
100 and 1000
200 and 2000

Span Limits CodeMPa ΔP

psi ΔP

B
C
D
E-1.54 and +2.4
-3.1 and +5.0
-7.7 and +10.0-14.0 and +20.0-220 and +350-440 and +750
-1100 and +1500-2000 and +3000

Range Limits (a), (b)(a) Negative values indicate that the higher of the two
measured pressures is on the normal “low side” (body side)of the transmitter. Positive values indicate higher pressure
is on the normal “high side” (connection block side). (b)Nonzero-based ranges require optional Zero Elevation orSuppression Kit.

Maximum Static Pressure:
The lower of the two measured pressures may
not exceed the following pressure limits:
Span Limits Code

MPa

psi

B
C
D
E

2.4
5.0
10.0
14.0

350
750
1500
2000

Ambient Temperature Limits:-40 and +120°C (-40 and +250°F).
Output Signal and Supply Pressure:Output Signal

Supply Pressure

20 to 100 kPa
3 to 15 psi

140 kPa
20 psi

Performance Specifications
Accuracy:±0.5% of calibrated span.
Repeatability:
0.1% of calibrated span.

Hysteresis:
Span Limits codes B, C, and D: 0.1% ofcalibrated span.
Span Limits code E: 0.2% of calibrated span.

Page 7

Pneumatic Transmitters (Differential Pressure)
11DM

Physical Specifications
Materials of Wetted Parts:

Enclosure Classification:

316 ss = AISI Type 316 stainless steel.
316L ss = AISI Type 316L stainless steel
Body
Capsule Bellows
Other Capsule Parts
Force Bar
Force Bar Seal
Force Bar Gasket
Capsule Gasket

Meets IEC IP53 and provides the environmental
protection of NEMA Type 3.

316 ss
316L ss or Monel
316 ss or Monel
316 sscobalt-nickel
silicone elastomer
silicone elastomer

Body Bolts:

Mounting:

Bracket for DN 50 or 2 in vertical or horizontal
pipe.
Air Connections:

The supply and output connections are tappedfor 1/4 NPT.

Alloy steel per ASTM A193 Grade B7.
Cover:High impact, glass filled polycarbonate.

Optional Features
Oxygen Service Preparation:

Cleaned and packaged for oxygen service.

NACE:

Compliance to NACE Standard MR-01-71.

How To Order
1) SPECIFY MODEL NUMBER: 11DMSpan Limits:___ ___		___ - ___

0.14 and 1.4 MPa or 20 and 200 psi........................................................................... B
0.3 and 2.8 MPa or 40 and 400 psi.............................................................................C
0.7 and 7.0 MPa or 100 and 1000 psi.........................................................................D
1.4 and 14 MPa or 200 and 2000 psi..........................................................................E

Wetted Parts:

316ss .......................................................................................................................................S316 ss except Monel bellows assembly.............................................................................. D

Process Connection:

1/4 NPT......................................................................................................................................................1
1/2 NPT......................................................................................................................................................2
R1/4 (metric).............................................................................................................................................3
R 1/2 (metric.............................................................................................................................................4

Optional Features: (also see previous page)

Zero elevation................................................................................................................................................... L
Zero suppression............................................................................................................................................. R

2) SPECIFY OUTPUT SIGNAL__________________________________________________________
3) SPECIFY CALIBRATED PRESSURE RANGE___________________________________________
4) SPECIFY OPTIONAL FEATURES_____________________________________________________
5) SPECIFY INFORMATION FOR INSTRUMENT TAG____________________________________

Page 8

Pneumatic Transmitters (Flow)
14A Series Pneumatic
Flow Integrators

14A

n Durable• Self-cleaning, self-aligning ball and thrust		 plate bearings• Gasketed cast aluminum covern Versatile• Field or panel mounting up to several
		 hundred feet from point of measurementn Stable• Force balance unit minimizes variations in		air supply pressure or ambient temperaturen Multiple Ranges• Changeable counting speed can match		 almost any flow range• No recalibration required

The 14A Series Pneumatic Flow Integrator
receives signal from transmitter, extracts
square root, and totalizes flow on integral
counter in engineering units.

For complete specifications, refer to GeneralSpecification Sheet GS 2B-5A1 A.

Functional Specifications
Input Signal and Suppiy Pressure
Intput Signal

Supply Pressure

20 to 100 kPa
3 to 15 psi

140 kPa
20 psiCounter:6-digit, nonreset.
Count Rates and Integrator Factors:
Count rate upper range values are shown in thefollowing table. Integrated flow is displayed directly when the count rate is set equal to the maximum
flow rate. Other flow rates can be accommodated
using the count rate times the integrator factor.
Example: Reading x 100 = Gallons.
Available Count Rate:
Upper Range Values
Counts per Minute
4.75 to 49.21
excluding:
5.46 to 5.54,
9.11 to 9.50,
and10.93 to 11.08

Counts per Hour
285 to 2953
excluding:
328 to 332,
547 to 570,and656 to 665

Counts per Day
6840 to 70870
excluding:
7870 to 7980,
13120 to 13680,
and
15750 to 15960

Page 9Pneumatic Transmitters (Flow)
14A

Performance Specifications
Accuracy:±0.5% of full scale calibrated between 30 and
100% of flow.

Input Signal:
Input cutoff point set for approximately 7% flow.
Input startup point set for approximately 11 % flow.

Repeatability:
0.2% of full scale.

Physical Specifications
Body:
Cast aluminum with acrylic enamel finish.
Specification

Cover		 High-impact, glass14A Seriesfilled polycarbonate,		gasketed, weatherproof.

Case—

Optional Features
Foxboro and CSA Certified:
For use in Class I, Groups A, B. C, and D,Division 2. For EP Series only.

Page 10Integral Mounting to d/p Cell Transmitter:For 14A-F. Supply regulator included withthis option.

Pneumatic Transmitters (Flow)
14A

How to Order
1) SPECIFY MODEL NUMBER: 14AIntegral Nonreset Counter:_________

Panel-Mounted.............................................................................................................................................—
Field-Mounted..............................................................................................................................................F
Field-Mounted with supply regulator.......................................................................................................FR

3) SPECIFY INPUT SIGNAL____________________________________________________________
4)	SPECIFY COUNTS PER MINUTE, HOUR, OR DAY(AT MAXIMUM FLOW RATE)_______________________________________________________
5) SPECIFY INTEGRATOR FACTOR____________________________________________________
6) SPECIFY INFORMATION FOR INSTRUMENT TAG_____________________________________

Page 11

Pneumatic Transmitters (Pressure)
11GM, 11GH Series
Pneumatic Gauge
Pressure Transmitters

11GM, 11GH

n 11GM—pressure measurement ranges

adjustable from 0-0.07 to
0­-14 MPa (0-10 to 0-2000 psi)

n 11GH—pressure measurement ranges

adjustable from 0-7.0 to 0-­80 MPa(0-1000 to 0-12000 psi)n Low Air Consumption

The 11GM, 11GH Series Pneumatic Gauge Pressure
Transmitters measure gauge pressure and transmit
a proportional pneumatic output signal.

For complete specifications, refer to
Product Specification Sheet PSS 2B-1A3 A.

Functional Specifications
Span, Range, and Overrange Limits:				SeriesSpan
Limits
Code

MPa

		
		
11GM
		
		

A
B
C
D
E

0.07 and 0.56
0.14 and 1.4
0.3 and 2.8
0.7 and 7
1.4 and 14

11GH
		

K
N

7 and 40
14 and 80				Seriespsi				SeriesSpan
Limits
Code

MPa

psi

10 and 80
20 and 200
40 and 400
100 and 1000
200 and 2000

		
		
11GM
		
		

A
B
C
D
E

0.7
3.5
7
14
28

100
500
1000
2000
4000

1000 and 6000
2000 and 12000

11GH
		

K
N

62
124

9000
18000

Span Limits

Maximum Overrange Pressure Limit (b)(b) Overrange limit of Span Limits code E with Wetted Parts
code M on Page 13 is reduced to 21 MPa (3000 psi).

Span
Limits
Code

MPa

psi

		
		
11GM
		
		

A
B
C
D
E

- 0.1 and +0.6
- 0.1 and +2.4
- 0.1 and +5- 0.1 and +10- 0.1 and +20-15 and +90-15 and +350
-15 and +750
-15 and +1500-15 and +3000

11GH
		

K
N

- 0.1 and +40- 0.1 and +80-15 and +6000-15 and +12000

Range Limits (a)(a) Nonzero-based ranges require optional zero elevation or
suppression kit. See “HOW TO ORDER.”

Output Signal and Suppiy Pressure
Output Signal

Supply Pressure

20 to 100 kPa
3 to 15 psi

140 kPa
20 psiAmbient Temperature Limits:-40 and +120°C (-40 and +250°F).
Body Temperature Limits:
11GM Series: -40 and +190°C (-40 and + 375°F).
11GH Series: -40 and +120°C (-40 and +250°F).

Performance Specifications
Repeatability:

Accuracy:		SeriesSpans Between
MPa		 psi		Accuracy(% of Span)All Spans

11GM Series:±0.5

Wetted Parts code S: 0.1% of calibrated span.
Wetted Parts code M: 0.2% of calibrated span.

11GH-K		

7 and 2020 and 40

1000 and 3000
3000 and 6000

±0.5
±1

11GH Series:

11GH-N		

14 and 40
40 and 80

2000 and 6000
6000 and 12000

±0.5
±1

11GM

Page 12

0.15% of calibrated span.

Pneumatic Transmitters (Pressure)
11GM, 11GH

Physical Specifications
Materials of Wetted Parts:
			
Item
Element

S
316ss

Materials:
Cover: High-impact, glass-filled polycarbonate.

Wetted Parts Code
M
Monel

C
Ni-SpanBody Bolts: Alloy steel per ASTM A193 Grade B7.

Connection
316ss
Monel
Block			316ssConnection
Gasket(none)

Enclosure Classification:
Meets IEC IP53 and provides the environmental
protection of NEMA Type 3.

Silicone
Viton-A
Elastomer		Air Connections:
The supply and output connections are tappedfor 1/ 4 NPT.

Optional Features
Air Supply Set:
Adjustable regulators with or without gauge.

High Accuracy Calibration (11GM only):
Accuracy of +0.25% of calibrated span.

Pressure Seals:
Refer to Page 34.

NACE (11 GM only):
Compliance to NACE Standard MR-01-75.

Oxygen Service Preparation:
Cleaned and packaged for oxygen service.

How to Order
1) SPECIFY MODEL NUMBER: 11GM-			 ___ ___ ___ - ___Span Limits:
0.07 and 0.56 MPa or 10 and 80 psi................................................................................... A
0.14 and 1.4 MPa or 20 and 200 psi................................................................................... B
0.3 and 2.8 MPa or 40 and 400 psi..................................................................................... C
0.7 and 7 MPa or 100 and 1000 psi.................................................................................... D
1.4 and 14 MPa or 200 and 2000 psi.................................................................................. E

Wetted Parts:

316ss ............................................................................................................................................... S
Monel (a)..........................................................................................................................................M

Process Connection:

1/4 NPT......................................................................................................................................................1
1/2 NPT......................................................................................................................................................2
R1/4 (metric).............................................................................................................................................3
R1/2 (metric).............................................................................................................................................4
Machined for 9/16-18 Aminco fitting (b)..............................................................................................5

Optional Features: (also see above)

Zero elevation................................................................................................................................................... L
Zero suppression............................................................................................................................................. R(a) Not available with Span Limits code A above.
(b) Not available with Span Limits code A or Wetted Parts code M above.

HOW TO ORDER Continued on Next Page.

Page 13

Pneumatic Transmitters (Pressure)
11GM, 11GH

How to Order (continued)
1) SPECIFY MODEL NUMBER: 11GH-			 ___ ___						
C ___ - ___Span Limits:

7 and 40 MPa or 1000 and 6000 psi................................................................................... K
14 and 80 MPa or 2000 and 12000 psi............................................................................... N

Wetted Parts:

Ni-Span element with 316 ss connection....................................................................................C

Process Connection:

1/2 NPT......................................................................................................................................................2
R 1/2 (metric)............................................................................................................................................4
Machined for 9/16-18 Aminco fitting....................................................................................................5

Optional Features: (also see previous page)

Zero elevation................................................................................................................................................... L
Zero suppression............................................................................................................................................. R

2) SPECIFY OUTPUT SIGNAL__________________________________________________________
3) SPECIFY CALIBRATED PRESSURE RANGE___________________________________________
4) SPECIFY OPTIONAL FEATURES_____________________________________________________
5) SPECIFY INFORMATION FOR INSTRUMENT TAG____________________________________

Page 14

Pneumatic Transmitters (Pressure)
11AL, 11AM, 11AH Series
Pneumatic Absolute
Pressure Transmitters

11AL, 11AM, 11AH

n 11AL—adjustable for ranges from 0-1.3 to

0-5.3 kPa (0-10 to 040 mmHg) absoluten 11AM—adjustable for ranges from 0-5.3

to 0-200 kPa (0-40 to 01520 mmHg)
absolute

11AH

n 11AH—adjustable for ranges from 0-0.7

to 0-2.7 MPa (0-10 to 0400 psi) absoluten Low air consumption

11AM

For complete specifications, refer to
Product Specification Sheet PSS 2B-1A1 A.

11AL

The 11AL, 11AM, 11AH Series Pneumatic Absolute
Pressure Transmitters measure absolute pressure
and transmit a proportional pneumatic output signal.

Functional Specifications
Span, Range, and Overrange Limits (in absolute pressure units):				
SeriesSpan
Limits		CodeSpan Limits				
SeriesSpan
LimitsCode

Maximum Overrange Pressure Limit
MPa

psi

11AL

L

1.3 and 5.3kPa

10 and 40 mmHg

11AL

L

0.8

115

11AM
		

M
H

5.3 and 53 kPa
50 and 200 kPa

40 and 400 mmHg
375 and 1520 mmHg

11AM
		

M
H

0.8
1.1

115
165

		
11AH
		

A
B
C

0.07 and 0.55 MPa
0.14 and 1.4 MP
0.3 and 2.7 MP

10 and 80 psi
20 and 200 psi
40 and 400 psi

		
11AH
		

A
B
C

0.7
2.4
5.2

100
350
750

		
		
SeriesMaximum Process Temperature:
120°C (250°F) at capsule.

Span
Limits		 Range Limits (a)
Code

11AL

L

0 and 9.3 kPa

0 and 70 mmHg

11AM
		

M
H

0 and 53 kPa
0 and 200 kPa

0 and 400 mmHg
0 and 1520 mmHg

		
11AH
		

A
B
C

0 and 0.62 MPa
0 and 2.4 MPa
0 and 5 MPa

0 and 90 psi
0 and 350 psi
0 and 750 psi(a) Nonzero-based ranges require optional zero suppression
kit (standard for 11AM). See “HOW TO ORDER.”

Ambient Temperature Limits:-40 and + 120°C (-40 and +250°F).
Output Signal and Supply Pressure:Output Signal

Supply Pressure

20 to 100 kPa
3 to 15 psi

140 kPa
20 psi

Performance Specifications
Accuracy:		Series11AL

Spans Between
kPa
mmHg
All Spans

Accuracy(% of Span)±1.0%

11AM-M		

5.3 and 13
13 and 53

40 and 100
1100 and 400

±0.5%
±1.0%

11 AM-H		

50 and 113
113 and 200|375 and 850
1850 and 1520

±0.5%
±1.0%

11AH

All Spans±0.5%

Repeatability:
11AL Series: 0.5% of calibrated span.
11AM, 11AH Series: 0.1% of calibrated span.
Hysteresis:
11AL Series: 0.5% of calibrated span.
11AM Series: 0.25% of calibrated span.
11AH Series: 0.15% of calibrated span.

Page 15

Pneumatic Transmitters (Pressure)
11AL, 11AM, 11AH

Physical Specifications
Materials:
Body Bolting: Alloy steel per ASTM A193 Grade B7.

Materials of Wetted Parts:
		
Wetted Part
Capsule Diaphragm
Other Capsule Parts
Force Bar
Force Bar Seal
Force Bar Gasket
Capsule Gasket
Process Connector Gasket

Body Material
AISI 316 ss
AISI 316L ss
316 ss
316 sscobalt-nickel
silicone elastomer
316 ss (a)ptfe (a)

Monel
Monel
Monel
Monel
cobalt-nickel
Viton-AMonelptfe(a) 11AH Series uses single silicone elastomer gasket as
capsule and connector gasket.

Cover: High-impact, glass-filled polycarbonate.
Mounting:
Bracket for nominal DN 50 or 2 in pipe.
Enclosure Classification:
Meets IEC IP53 and provides the protection of
NEMA Type 3.
Air Connections:
The supply and output connections are tappedfor 1/4 NPT.

Optional Features
Air Supply Sets:
Adjustable regulators with or without
output gauge.

Oxygen Service Preparation:
Cleaned and packaged for oxygen service.

How to Order
1) SPECIFY MODEL NUMBER: 11AL-	­				___								L ___S ___ – ___Span Limits:

1.3 and 5.3 kPa or 10 and 40 mmHg absolute.................................................................L

Body Material:

AISI Type 316 stainless steel (316 ss)..........................................................................................S

Process Connections:Tapped for 1/4 NPT............................................................................................................................... 1
Tapped for 1/2 NPT............................................................................................................................... 2
Tapped for R1/4 (metric)....................................................................................................................... 3
Tapped for R1/2 (metric)....................................................................................................................... 4
Welding neck for 14 x 21 mm tube and 1/2 in Schedule 80 pipe.................................................. 6

Optional Zero Suppression...........................................................................................................................R
HOW TO ORDER Continued on Next Page.

Page 16

Pneumatic Transmitters (Pressure)
11AL, 11AM, 11AH

How to Order
1) SPECIFY MODEL NUMBER: 11AM -	­				___ ___ ___Span Limits:

5.3 and 53 kPa or 40 and 400 mmHg absolute............................................................... M
50 and 200 kPa or 375 and 1520 mmHg absolute......................................................... H

Body Material:

316ss ............................................................................................................................................. S
Monel Not available with Span Limits code H above..............................................................M

Process Connections:Tapped for 1/4 NPT............................................................................................................................... 1
Tapped for 1/2 NPT............................................................................................................................... 2
Tapped for R1/4 (metric)....................................................................................................................... 3
Tapped for R1/2 (metric)....................................................................................................................... 4
Welding neck for 14 x 21 mm tube and 1/2 in Schedule 80 pipe.................................................. 6

1) SPECIFY MODEL NUMBER: 11AH-	­				___ ___									S ___ – ___Span Limits:

0.07 and 0.55 MPa or 10 and 80 psi absolute................................................................. A
0.14 and 1.4 MPa or 20 and 200 psi absolute..................................................................B
0.3 and 2.7 MPa or 40 and 400 psi absolute................................................................... C

Body Material:

316 ss ............................................................................................................................................. S

Process Connections:Tapped for 1/2 NPT............................................................................................................................... 2
Tapped for R1/2 (metric)....................................................................................................................... 4

Optional Zero Suppression...........................................................................................................................R2) SPECIFY OUTPUT SIGNAL__________________________________________________________
3) SPECIFY CALIBRATED ABSOLUTE PRESSURE RANGE________________________________
4) SPECIFY OPTIONAL FEATURES_____________________________________________________
5) SPECIFY INFORMATION FOR INSTRUMENT TAG____________________________________

Page 17

Pneumatic Transmitters (Repeater)
139PP Pneumatic Pressure
Repeater

139PP

For transmitting the pressure of corrosive, hot,gaseous, or liquid media or other fluids tending
to solidify or clot. The pressure exerted by themedium is converted in a 1:1 ratio into a pneumaticpressure. For the repeater with zero suppression,
the output is equal input plus suppression.
For complete specification, refer to ProductSpecification Sheet PSS EMP0140 A-(en)n Replacement for 17R
n Adjustable up to 10 bar
n Overrange protection up to 16 barn Vacuum proofn With damping facilitiesn Version with exchangeable measuringcapsule, extended diaphragmn Version with welded in measuringcapsule, process side without gap,
flush diaphragmn Version for zero suppression to –0,5 barn Version with wetted parts in Tantalumn PTFE diaphragm protector

Functional Specifications
Input
Process pressure........ 0.01 to 10 bar
Repeater withzero suppression....... initial value adjustablefrom –0.05 to –0.5 bar
Overload limits........... –1 bar and 16 bar
PTFE diaphragm
protector..................... 0 bar and 16 bar, notsuitable for vacuum service

Error Influence
Air supply fluctuation............. ≤ 1 % / 0.1 bar
Ambient and processtemperature fluctuation:
Repeater without
zero suppression.................... ≤ 0.5 mbar / 10 K
with zero suppression............ ≤ 1 mbar / 10 K
Tantalum version.................... ≤ 2 mbar / 10 K
PTFE diaphragm protector.... ≤ 1 mbar / 10 K

Ambient conditions
Operating temperature and application class:Repeater without
zero suppression......... –25 to +200 °C
with zero suppression.–25 to +125 °C
Tantalum version......... –25 to +155 °C
PTFE diaphragm.......... –25 to +100 °C
Storage temperature... –40 to +125 °C
Humidity........................ ≤ 100 %, condensation
permitted
Protection class............ IP 54, IP 66

Performance Specifications
Accuracy:
Output pressure up to

2 bar

3 bar

4 bar

5 bar

Repeater without zero suppression
and Tantalum version(mbar)

≤1.5

≤3

≤6

≤12

≤20

Repeater with zero suppression and
Repeater with PTFE diaphragm protector(mbar)

≤3

≤4

≤7

≤12

≤20

Repeater without zero suppression(mbar)

<0.1

<0.2

<0.3

<0.5

<1

Repeater with zero suppression(mbar)

<0.2

<0.3

<0.4

<0.6

<1

Tantalum version and
Repeater with PTFE diaphragm protector(mbar)

≤0.5

≤0.6

≤0.7

≤1

≤1.5

Inherent Air Consumption

(l/h)

120

160

210

260

360

Air Supply

(bar)

1.4

2.5

4.5

6.5

11

Accuracy

Hysteresis

Page 18

1 bar

Pneumatic Transmitters (Repeater)
139PP

Application

The repeater is usually used in closed-tankliquid-level applications in conjunction
with a differential pressure liquidlevel transmitter such as the 13F(pneumatic), or the IP10 (electronic).

How to Order
Model Codes
Pneumatic Pressure Repeater

139PP		

						

Initial Value:

0.01 bar 0.15 psi.................................................................................................. -A- 0.5 bar -7.5 psi (a).............................................................................................. -B

Wetted parts material:

1.4571 (316Ti)................................................................................................................ S
Tantalum......................................................................................................................... T

Version:Extended diaphragm (capsule screwed in) (not available with Wetted Parts T)........... E
Flush diaphragm (capsule welded in) (b)........................................................................... FMounting Orientation:

horizontal, process side above.....................................................................................................A
horizontal, process side below...................................................................................................... B
diaphragm vertical..........................................................................................................................C

Output Pressure:

up to 1 bar................................................................................................................................................1
up to 2 bar................................................................................................................................................2
up to 4 bar................................................................................................................................................3
up to 6 bar................................................................................................................................................4
up to 10 bar..............................................................................................................................................5

Optionals:

PTFE-foil with silicone oil (only for version F) (c)........................................................................................ -P
Certificate EN 10204-2.1 (DIN 50 049-2.1).................................................................................................. -1
Certificate EN 10204-2.3 (DIN 50 049-2.3).................................................................................................. -2
Tag.No. Labeling stainless steel label fixed with wire............................................................................... -L
Tag.No. Labeling stamped with weather-resistant color........................................................................... -S
31A = 3 inch Class 160 ANSI 4 holes (to repace 17R from Foxboro)..................................................... 31A		
33D= DN80, PN 10/40 8 holes (to repace 17R from Foxboro)...............................................................33O

Notes(a) Output = Input + 0.5 bar(b) Suitable for food industry(c) Not for vacuum(e) Not available with Option-P(d) Not available with Material T, not for Version E

Page 19

Pneumatic Transmitters (Liquid Level and Density)
Pneumatic Liquid Level
or Density Flanged
d/p Cell Transmitters

15F, 13F, 13FE

n 15F — Low Range Flush Diaphram

Transmitter adjustable for ranges from
0-0.13 to 0-0.63 m (0-5 to 0-25 in) headof watern 13F — Flush Diaphram Transmitter

adjustable for ranges from 0-0.51 to
0-21.6 m (0-20 to 0-850 in) head ofwater

n 13FE — Extended Diaphram Transmitterfor measurements from 0-0.51 to 0-21.6
m (0-20 to 0-850 in) head of water

The 15F, 13F, 13FE Series Pneumatic
Liquid Level or Density Flanged d/p Cell
Transmitters measure hydrostatic headof liquid and transmit a proportional
pneumatic output signal.

n Low Air Consumption

For complete specifications, refer to
Product Specification Sheet PSS 2B- 1D2 A.

Functional Specifications
Span Limits:				SeriesSpan
LimitsCodemetersinches15FL

0.13 and 0.63

5 and 25

13F,
13FE

M
H

0.51 and 6.3
5.1 and 21.6

20 and 250
200 and 850

Span Limits (Head of Water)

Range Limits:				SeriesSpan
Limits
Code

15F
13F,
13FE

Range Limits (Head of Water) (a)metersinches

L

±1.3

±50

M
H

±6.3
±21.6

±250
±850

Process Temperature Limits:
ANSI Flanges: -40 and +190°C (-40 and +375°F).
DIN Flanges:-40 and +120°C (-40 and +250°F).
Ambient Temperature Limits:-40 and +120°C (-40 and +250°F).
Output Signal and Supply Pressure:Output Signal

Supply Pressure

20 to 100 kPa
3 to 15 psi

140 kPa
20 psi(a) Non zero-based ranges require optional zero elevationor suppression kit. Upper and lower range values must notexceed range limits. Negative numbers indicate a higher
pressure on the normal “low side” of the transmitter, such as
may occur in a closed tank application with a wet leg.

Performance Specifications
Accuracy:±0.5% of calibrated span for all models except
for spans greater than 13.3 m (525 in) head ofwater which are ±0.75%.
Repeatability:
0.1% of calibrated span.

Page 20

Hysteresis:
15F Series: 0.1 % of calibrated span or 0.25 mm(0.01 in) head of water, whichever is larger.
13F and 13FE Series: 0.1 % of calibrated spanor 1.25 mm (0.05 in) head of water, whichever islarger.

Pneumatic Transmitters (Liquid Level and Density)
15F, 13F, 13FE

Physical Specifications
Body and Flange Material:
Zinc cobalt-plated carbon steel or 316 ss (b), as
specified.
Materials of Wetted Parts:
High Pressure Side:
Capsule Diaphragm: 316L ss (c).
Other Capsule Parts: 316 ss.
Capsule Gaskets:
		 15F,13F Series: 316 ss.
		 13FE Series: None.
Retaining Ring (d): 316 ss.
Low Pressure Side (Partial List):
Capsule Diaphragm: 316L ss.
Force Bar: 316 ss.
Force Bar Seal: Cobalt-nickel.
Force Bar Gasket: Silicone elastomer.
Capsule Gaskets:
		 15F, 13F Series: 316 ss.
		 13FE Series: ptfe.

Mounting:
Provided by process connection flange. See“HOW TO ORDER.”

Nominal Diaphragm Extension Length “A”(13FE Series): See “HOW TO ORDER.”
Flange Extension Length “B”:
15F and 13FE Series: 125 mm (5 in).
13F Series: 125 or 200 mm (5 or 8 in), as specified.
Enclosure Classification:
Meets IEC IP53 and provides the environmental
protection of NEMA Type 3.

(b) AISI Type 316 stainless steel.
(c) AISI Type 316L stainless steel.
(d) Raised face portion of flange.

Cover Material:
High-impact, glass-filled polycarbonate.

Optional Specifications
Wetted Parts Materials:
Monel, Duranickel, Hastelloy C, tantalum, andvarious coatings.
High Process Temperature:
To 315°C (600°F).

Lower Spans:
Provides spans as low as one-half standard.
Air Supply Sets:
Adjustable regulators with or without gauge.

How to Order
1) SPECIFY MODEL NUMBER: 15F-	­		- L		5 Transmitter:
316 ss body and flange...........................................................................................A
Zinc cobalt-plated carbon steel body and flange..............................................A1

Span Limits:

0.13 and 0.63 m or 5 and 25 in head of water.............................................................L

Body and Flange Material; and Connection:

Zinc cobalt-plated carbon steel 6 in ANSI Class 150.......................................................... K61A
316 ss		
6 in ANSI Class 150...........................................................S61A
Zinc cobalt-plated carbon steel 150 mm,16 Bar, DIN, and BS.......................................... K61D
316 ss		
150 mm,16 Bar, DIN, and BS.......................................... S61D

Flange Extension Length:

125 mm (5 in)............................................................................................................................................ 5

Elevation or Suppression (Optional):

Zero elevation kit.............................................................................................................................................. L
Zero suppression kit........................................................................................................................................ R

HOW TO ORDER Continued on Next Page.

Page 21

Pneumatic Transmitters (Liquid Level and Density)
15F, 13F, 13FE

How to Order (continued)
1) SPECIFY MODEL NUMBER: 13F-	­Transmitter:___ - ___ ______ ___ - ___316 ss body and flange........................................................................................... A
Zinc cobalt-plated carbon steel body and flange.............................................. A1

Span Limits:

0.51 and 6.3 m or 20 and 250 in head of water.......................................................... M
5.1 and 21.6 m or 200 and 850 in head of water.........................................................H

Body and Flange Material; and Connection:

Carbon steel 3 in ANSI Class 150..........................................................................................K31A
Carbon steel 3 in ANSI Class 300......................................................................................... K33A
316 ss		
3 in ANSI Class 150..........................................................................................S31A
316 ss		
3 in ANSI Class 300..........................................................................................S33A
Carbon steel 80 mm,10/40 Bar, DIN, and BS...................................................................... K33D
316 ss		
80 mm,10/40 Bar, DIN, and BS...................................................................... S33D

Flange Extension Length:

125mm(5in)............................................................................................................................................... 5
200 mm (8 in)............................................................................................................................................ 8

Elevation or Suppression (Optional):

Zero elevation kit.............................................................................................................................................. L
Zero suppression kit........................................................................................................................................ R

1) SPECIFY MODEL NUMBER: 13FE-	­		-			Transmitter:5		-

316 ss body and flange.................................................................................. A
Zinc cobalt-plated carbon steel body and flange..................................... A1

Span Limits:

0.51 and 6.3 m or 20 and 250 in head of water..................................................M
5.1 and 21.6 m or 200 and 850 in head of water................................................ H

Body and Flange Material; and Connection:
Zinc cobalt-plated carbon steel
Zinc cobalt-plated carbon steel
316 ss		
316 ss		Zinc cobalt-plated carbon steel
Zinc cobalt-plated carbon steel
316 ss		
316 ss		

4 in ANSI Class 150....................................................K41A
4 in ANSI Class 300....................................................K43A
4 in ANSI Class 150....................................................S41A
4 in ANSI Class 300....................................................S43A
100 mm,16 Bar, DIN, and BS....................................K41D
100 mm, 40 Bar, DIN, and BS...................................K43D
100 mm,16 Bar, DIN, and BS....................................S41D
100 mm, 40 Bar, DIN, and BS...................................S43D

Flange Extension Length:

125 mm (5 in)................................................................................................................................... 5

Nominal Diaphragm Extension Length:

50 mm (2 in)..............................................................................................................................................2
100 mm (4 in)............................................................................................................................................4
150 mm (6 in)............................................................................................................................................6

Elevation or Suppression (Optional):

Zero elevation kit.............................................................................................................................................. L
Zero suppression kit........................................................................................................................................ R

2) SPECIFY OUTPUT SIGNAL__________________________________________________________
3) SPECIFY CALIBRATED DIFFERENTIAL PRESSURE RANGE_____________________________
4) SPECIFY OPTIONAL FEATURES_____________________________________________________
5) SPECIFY INFORMATION FOR INSTRUMENT TAG____________________________________

Page 22

Pneumatic Transmitters (Liquid Level)
13DMP, 13DEMP Series
Pneumatic d/p Cell
Transmitters With Matched
Pressure Seals for Liquid
Level Measurement

13DMP, 13DEMP

n Low Air Consumption
n 13DMP—FLUSH TYPE with 3-inch flange

connections

n 13DEMP—EXTENDED TYPE with 4-inchflange connectionsn Pressure Seals• Eliminate problems caused by repeaters		 and wet and dry legs
• Make installation and adjustments easier		 than other approaches• Provide positive isolation with a minimum
		 of process wetted parts• Allow higher process temperaturesn Ideal for Pressurized or Evacuated

The 13DMP, 13DEMP Series Pneumatic d/p
Cell Transmitters with matched pressure seals
measure differential pressure and transmit a
proportional pneumatic output signal.

Vessels

For complete specifications, refer to ProductSpecification Sheet PSS 2B-1C1 B.

Functional Specifications
Span and Range Limits:
Span Limits (Head of Water)
Seriesmetersinches13DMP-M,13DEMP-M
13DMP-H,13DEMP-H

0.51 and 6.4
5.1 and 21.6

20 and 250
200 and 850

Range Limits (Head of Water) (a)Seriesmetersinches13DMP-M,13DEMP-M
13DMP-H,13DEMP-H- 6.4 and +6.4
-21.6 and +21.6-250 and +250
-850 and +850(a) Negative numbers indicate a higher pressure on thenormal “Low Side” of the transmitter.

Process Temperature Limits:-37 and +150°C (-35 and +300°F).
Ambient Temperature Limits:-37 and +80°C (-35 and +180°F).
Output Signal and Supply Pressure:Output Signal

Supply Pressure

20 to 100 kPa
3 to 15 psi

140 kPa
20 psi

Performance Specifications
Accuracy:
Spans (Head of Water) Between			metersinches0.51 and 13
13 and 21.6

20 and 525
525 and 850

Accuracy (b)% of Span
±0.5
±1.0

Repeatability:
0.1% of calibrated span.
Hysteresis:
0.2% of calibrated span.

(b) Includes linearity, hysteresis, and repeatability.

Page 23

Pneumatic Transmitters (Liquid Level)
13DMP, 13DEMP

Physical Specifications
Materials of Wetted Parts:
Remote Seal
Diaphragms

Raised Face Backup
Pads (Gasket Surface)

AISI Type 316L
stainless steel(316L ss)AISI Type 316
stainless steel(316 ss)

Hastelloy C276

Hastelloy C
Tantalum

Materials of Nonwetted Parts:
Description

Material

Flanges
Capillary
Transmitter Body
Fill Fluid

316 ss or carbon steel
316ss
316ss
DC 200 silicone (c)(c) Relative density (specific gravity) is 0.90 at 25°C (77°F).

Capillary Lengths:
See “HOW TO ORDER.”
Mounting:
Seals mount by process connection flanges. See“HOW TO ORDER.” Transmitter body mounts tobracket for DN 50 or 2 in vertical or horizontal
pipe.
Enclosure Classification:
Meets IEC IP53 and provides the environmental
protection of NEMA Type 3.
Cover Material:
High-impact, glass-filled polycarbonate.
Air Connections:
The supply and output connections are tappedfor 1/4 NPT.

Optional Specifications
Air Supply Set:
Adjustable regulators with or without gauge.

High Temperature Fill:
For process temperatures up to 300°C (580°F).

How to Order
1) SPECIFY MODEL NUMBER: 13DMP-___ ___ ___ ______

Span Limits:

0.51 and 6.4 m or 20 and 250 in head of water.......................................................... M
5.1 and 21.6 m or 200 and 850 in head of water........................................................ H

Wetted Parts Material:

316 ss with 316L ss diaphragm...............................................................................................S
Hastelloy C with Hastelloy C276 diaphragm........................................................................ C
Tantalum.....................................................................................................................................T

Capillary Length (each capillary):

1.5m( 5ft)............................................................................................................................................ A
4.5m(15ft)........................................................................................................................................... B
8 m (26 ft)........................................................................................................................................... C

Process Connections (nonprocess wetted):
Zinc cobalt-plated carbon steel
Zinc cobalt-plated carbon steel
Zinc cobalt-plated carbon steel
316 ss		
316 ss		
316 ss		Zinc cobalt-plated carbon steel
316 ss		

HOW TO ORDER Continued on Next Page.

Page 243 in ANSI Class 150........................................................................K31A
3 in ANSI Class 300........................................................................K33A
3 in ANSI Class 600........................................................................K36A
3 in ANSI Class 150........................................................................S31A
3 in ANSI Class 300........................................................................S33A
3 in ANSI Class 600........................................................................S36A
80 mm 10/40 Bar, DIN and BS.....................................................K33D
80 mm 10/40 Bar, DIN and BS.....................................................S33D

Pneumatic Transmitters (Liquid Level)
13DMP, 13DEMP

How to Order (continued)
1) SPECIFY MODEL NUMBER: 13DEMP-	­___ ___ ___ ___ ______
Span Limits:

0.51 and 6.4 m or 20 and 250 in head of water................................................. M
5.1 and 21.6 m or 200 and 850 in head of water................................................H

Wetted Parts Material:

316 ss with 316L ss diaphragm.......................................................................................S
Hastelloy C with Hastelloy C276 diaphragm................................................................ C

Capillary Length (each capillary):

1.5m (5 ft).................................................................................................................................. A
4.5m (15 ft)................................................................................................................................ B
8m (26 ft)................................................................................................................................... C

Extension Length:

50mm (2 in).........................................................................................................................................2
100mm (4 in).......................................................................................................................................4150mm (6 in).......................................................................................................................................6

Process Connections (nonprocess wetted): Carbon steel, 4 in ANSIClass 150.....................................................................................................................................................K41A
Class 300.....................................................................................................................................................K43A
Class 600.....................................................................................................................................................K46A

2) SPECIFY OUTPUT SIGNAL__________________________________________________________
3) SPECIFY CALIBRATED DIFFERENTIAL PRESSURE RANGE_____________________________
4) SPECIFY OPTIONAL FEATURES_____________________________________________________
5) SPECIFY INFORMATION FOR INSTRUMENT TAG____________________________________
Note: For liquid level applications, zero elevation kits always required due to wet leg effect of filled seal
system; therefore, it is a standard part of the 13DMP and 13 DEMP Series Transmitters.

Page 25

Pneumatic Transmitter (Liquid Level and Density)
167LP Pneumatic Buoyancy
Transmitter with Torque
Tube for Liquid Level,
Interface and Density

167LP

n Level transmission between vessel andtransmitter by torque tuben Applicable for service temperatures from

This transmitter is designed to performmeasurements for liquid level, interface anddensity of liquids. The measurement is basedon the Archimedes buoyancy principle.
For complete specification, refer to ProductSpecification Sheet PSS EML0110 A-(en)–196°C to +400°C and pressures up to
PN 250n The span can be set over a 1:5 ration A wide selection of materials facilitates
service under corrosive conditionsn Material opproval certificates to EN
10204-3.1 available
n Various licences in accordance withnational regulationsn Licensed for use on sea ships in theGermanische Lloyd class, or on otherstructures classified by Germanische Lloyd

Functional Specifications
Input
Measuring span...........3 to 15 N
Density range...............100 <ρ <1600 kg/m3
Standard lengths ofdisplacers 204DE........350 to 3000 mm
14 to 120 inch
Weight of displacer....max. 25 N
Output...........................0.2 to 1 bar/3 to 15 psi/20
to 100 kPa/0.2 to 1 kp/cm2
Supply air......................1.4 ±0.1 bar or 20 ±1.4 psi
Transitional Behavior
Relative error...................<1%
Sensitivity..........................<0.1%
Ambient temperature
influence...........................<0.2%/10 K
Process temperature
influence...........................<0.1%/10 K
Supply air influence........<0.2%/0.1 bar
Air consumption.............<200 l/h
Air capacity......................1200 l/h
Load effect(measured at 0.6 bar)...+3% for 400 l/h exhaused
flow -3% for 400 l/h
delivered flow

Page 26

Operating conditions
Process temperature-196°C to +400°C
Pressure ratingacc. to DIN.........................PN 16, 40, 63, 100, 160, 250
acc. to ANSI.......................Class 150, 300, 600, 900,1500
with heating jacket.............wafer body		max. PN 160/Class 900;		heating jacket PN 25,
		heating with saturated steam		or thermal oils
Ambient temperature.......-40 to +90°C
Relative humidity................<100%
Condensation..................... permitted
Transportation andstorage temperature.......-40 to +90°C
Protection class...................IP 55 (acc. to DIN 40 050)
The device can be operated at a class D2 location inaccordance with DIN IEC 654, part 1.
Mounting
Mounting method..............sandwich mounted
acc. to DIN.........................DN 80, DN 100
acc. to ANSI.......................3 inch, 4 inch
Pneum. connections..........internal thread		
DIN 45 141-Q 1⁄4-18 NPT

Pneumatic Transmitters (Liquid Level and Density)
167LP

How to Order

Specify model number 167LP		
Wafer Body with Indicator (Flange Size and Pressure Rating):						

DN80 PN16-40 (available with Contact Face C)................................................. -20
DN80 PN16-160 (available with Contact Face U, N ).........................................-21
DN80 PN16-250 (available with Contact Face E, L )...........................................-22
DN100 PN16-160 (available with Contact Face U, N ).......................................-23
DN100 PN16-250 (available with Contact Face E, L )........................................ -24
DN100 PN16-40 (available with Contact Face C).............................................. -25
3-Inch ANSI Class 150............................................................................................ -31
3-Inch ANSI Class 300/600/900.............................................................................-32
3-Inch ANSI Class 1500...........................................................................................-34
4-Inch ANSI Class 150............................................................................................ -41
4-Inch ANSI Class 300/600/900.............................................................................-42
4-Inch ANSI Class 1500...........................................................................................-44

Wafer Body Contact Face:

Type C/C Raised Face (Rz 40 - 160) Per DIN 2526 (available with -20, -25).......................................... CType E/E Raised Face (Rz 0 - 16) Per DIN 2526 (available with -22, -24)................................................E
Type N/F (Grove) Per DIN 2512 (available with -21, -23).......................................................................... U
Type N/N (Grove) Per DIN 2512 (available with -21, -23) ........................................................................ N
Type L/L DIN 2696 (available with -22, -24).................................................................................................. L
Type RF/RF Raised Face Per ANSI B16.5 (available with -31, -32, -34, -41, -42, -44)...........................RType RJF/RJF Ring Joint Face Per ANSI B16.5 (available with- 31, -32, -34, -41, -42, -44)................ J
Type SF/SF Smooth Finish (125 microinch) (available with -31, -32, -34, -41, -42, -44).......................S

Wafer Body Material (Process Wetted):

Carbon Steel 1.0460 (A-105)...................................................................................................... K
1.4404 (316L).................................................................................................................................S
Hastelloy C..................................................................................................................................... C

Wafer Body Mounting Direction (Transmitter on body):

Right Hand Mounted..............................................................................................................................R
Left Hand Mounted................................................................................................................................ L

Torque Tube Material (Process Wetted):

316 (1.4571/1.4404/1.4435)...........................................................................................................................S
Hastelloy C........................................................................................................................................................C
Inconel 600........................................................................................................................................................ I
Monel................................................................................................................................................................ M

Signal Range:

0.2 To 1.0 bar..............................................................................................................................................................1
3 To 15 psi...................................................................................................................................................................2
20 To 100 kPa..............................................................................................................................................................3
0.2 To 1.0 kp/cm2.......................................................................................................................................................4

Optional Features:

Oil Damping........................................................................................................................................................................ -D
Electrical Certificates:
ATEX — II 1/2 G c IIC (Zone 0) for media AI, AII, B(c)...................................................................................................... -E
ATEX — II 2 G c IIC (Zone 1) for media AI, AII, B............................................................................................................. -P
Overfill Protection Per WHG For Environmental Pollution Fluids(e)............................................................................ -V
Certificates:
EN 10204-2.1....................................................................................................................................................................... -1
EN 10204-2.2 Specific Test Report (Calibration)............................................................................................................ -2
EN 10204-3.1 Inspection Certificate of Process Wetted Metallic Material................................................................. -3
PED 97/23/EC additional unit verification, according to module F/G........................................................................ -4
Comply with NACE Standard MR-01-75 (available with Wafer Body Material Code S
and Torque Tube Material Code C, I or M only).......................................................................................................... -6
Material Test:
X-Ray And Isotope Test For Weldings............................................................................................................................. -7
Dye Penetrate Test............................................................................................................................................................. -8
Tag No. Labeling:
Stainless Steel Label Fixed With Wire...............................................................................................................................-L

Notes(c) Available with Contact Face E, N, R & S(e) Not available with Wafer Body -33, -34, -43 & -44

Page 27

Pneumatic Transmitters (Temperature)
P12A Series Pneumatic
Temperature Transmitters

12A

n Trouble-Free• Force balance mechanism has no moving
		partsn Accuraten Responsive• Comparable to an unsheathed thermocouple

n Simple Range Change• Span can be elevated or suppressed 100%
• Adjustment involves only three parts

n Easy Installation• A variety of mounting positions

The 12A Series Pneumatic Temperature
Transmitters measure temperature and transmit
a proportional pneumatic output signal.

n Low Air Consumption

For complete specifications, refer to ProductSpecification Sheet PSS 2B-1B1 A.

Functional Specifications
Range Limits, Spans, and Maximum Overrange Temperatures (Maximum ORT):					
Sensor
		 Sensitive		
Length “X”

Fahrenheit (F°)
Range Limits		

SeriesmminLower				 -100
12A*			
-100
and
150
6
-100
12AS*			-100				 -100

Upper
+330
+1000
+1000
+1000
+1000

Span
50
100
125
150
200

Maximum
ORT
330
1000
1000
1000
1000

				 -100
		
-100
75
3
				 -100
			 -100
12A
-100
and
75
3
-100
12AS
or
or
		
-100
150
6
				 -40

+150
+650
+850
+1000

50
100
150
200

330
1000
1000
1000

+1000
+1000
+1000
+1000

250
300
400
500

1000
1000
1000
1000

		
0
150
6
				 +200

+1000
+1000

600
800

1000
1000

				 -350				 -350				 -300
12A-C
150
6
-250
				 -200				 -150

+330
+400
+600
+600
+600
+600

50
100
150
200
250
300

330
600
600
600
600
600

-250

+600

200

600

+600
+500
+400
+300
+200

+1400
+1400
+1400
+1400
+1400

400
500
600
700
800

1400
1400
1400
1400
1400

		

75

3

				
				
12A-H
150
6
				
				

*See statement under “PERFORMANCE SPECIFICATIONS.”					
SensorCelsius (C°)
		 Sensitive		
Length “X”		
Range Limits
Maximum
SeriesmminLower Upper		
Span
ORT				 -75
12A*
			 -75
and
150
6
-75
12AS*
			 -75

+125
+460
+550
+550

25
50
75
100

125
460
550
550

				
		
75
3
12A			
and			

-75
-75
-75
-75

+70
+300
+550
+550

25
50
75
100

125
460
550
550

12AS
75 or
		 150

3 or
6

-75
-75

+550
+550

150
200

550
550

		 150

6

0

+550

300

550

				 -210				 -220
150
6
-170
12A-C
				 -140				 -95

+125
+200
+320
+320
+320

25
50
100
150
200

125
320
320
320
320

		

-160

+320

100

320

				 +350
12A-H
150
6
+250
				 +150

+760
+760
+760

200
300
400

760
760
760

75

3

Output Signal: 20 to 100 kPa or 3 to 15 psi.
Ambient Temperature Limits:-35 and +80=°C (-30 and +180°F). The vinyl
covered portion of the capillary tubing must bein an area between -55 and +105°C (-65 and+220°F).
Supply Pressure: 140 kPa or28 20 psi.

Page 28

Pneumatic Transmitters (Temperature)
12A

Performance Specifications(12A and 12AS Series with range limits, spans,and sensitive lengths as noted by asterisks in
Range Limits table.)
Accuracy (Includes linearity, hysteresis, andrepeatability): ±0.5% of calibrated span.

Repeatability:
0.2% of calibrated span.
Step Response Time (Basic thermal system andagitated water bath): Maximum of 12 seconds for a
90% recovery from a step change on a bare sensor
and less than 2 seconds for a 63% recovery.

Physical Specifications
Body:
Die-cast, low copper aluminum alloy with baked
gray vinyl finish.
Cover:
Blue, high impact, glass filled polycarbonate.
Environmental Protection:
The transmitter housing is weatherproof. It
is dust-protected as defined in IEC IP53 and,with its constant air purging, provides theenvironmental protection of NEMA Type 3.
Thermal System:
SAMA (Scientific Apparatus Makers’ Association)
Class IIIB gas pressure system.
Connecting Tubing:Vinyl covered AISI Type 302 stainless steel (302ss) flexible armor over 316 ss capillary. The
length is 1m (3.5 ft).
Bushings:
Fabricated of 316 ss and available with R1/2,
R3/4, R1, G1/2B, G3/4B, G1B, 1/2 NPT, 3/4 NPT,or 1 NPT external thread and 1-18 NS internal
thread. Bushing not required if temperature iswell specified.

Sensor:Fitted with an adjustable union and bendable
extension. The jam nut has an external 1-18
UNS thread. All process wetted parts are 316 ss
except for the 12A-H Series, which has a sensor
and bendable extension fabricated of Inconel.
Also, the 12A-H Series has a fixed union.
Diameter (Y): 9.53 mm (0.375 in).
Sensitive Length (X): 75 or 150 mm (3 or 6 in).
Extension Length (J): 450 mm (18 in). the J
dimension for the 12A-A Series must be longenough to limit the temperature of the capillarytubing to 105°C (220°F).
Insertion Length (U): 200 to 530 mm (8 to 21 in),adjustable.
Immersion Length (R): As specified.
Air Connections:
The supply and output connections are tappedfor 1/4 NPT.
Plain Bendable Sensor with Adjustable Union:

Adjustable Union Packing:
ptfe: For temperatures up to 205°C (400°F).
Graphite Impregnated Inconel Mesh: For
maximum temperature of 540°C (1000°F).

Optional Features
Air Supply Sets:
Available as fixed or adjustable filter regulatorswith or without gauges. Refer to Page 27.
Sausage Type Sensors:
For measurement of gas temperature in ducts orflues.
Thermowells:
For mechanical or corrosive protection. Refer toPage 35.

Derivative Unit:
Compensates for thermal lag in process.
Intermediate Spans:
Spans different from those listed in table under“FUNCTIONAL SPECIFICATIONS.”
Optional Thermal System Configurations:
Other connection tubing, sensor types, andmounting are available.

Page 29

Pneumatic Transmitters (Temperature)
12A

How to Order
1) SPECIFY MODEL NUMBER:________	­ – ______ ____ ___ ___ ___Transmitter with Basic Thermal System
and Code 3542 Sensor...................................................................... 12A
Cryogenic Thermal System
and Code 3542 Sensor.................................................................... 12A-C
High Temperature Thermal System
and Code 3482 Sensor.................................................................... 12A-H
Sanitary Thermal System(see additional specifications below).......................................... 12AS
ADDITIONAL SPECIFICATIONS FOR
12AS SERIES SANITARY THERMAL SYSTEMS ONLY:
Sensor Code:........................................................................................................... 3242
Connector:

16A Bevel Seat...................................................................................................................... S1
16AMP Clamp Seat............................................................................................................... S2Type RJT................................................................................................................................. S3Type ISS.................................................................................................................................. S4
ISO Clamp Flange................................................................................................................. S5
DIN Seat................................................................................................................................. S6

Nominal Line Size:

25 mm (1 in)..................................................................................................................................... A
40mm(1 1/2in)................................................................................................................................. B
50 mm (2 in)..................................................................................................................................... C
65mm(21/2in).................................................................................................................................. D
75 mm (3 in}..................................................................................................................................... E
100 mm (4 in)................................................................................................................................... F

Sensor Sensitive Length (X):

75mm (3 in)...............................................................................................................................................R
150 mm (6 in)............................................................................................................................................T

Capillary Tubing Length:

1 m (3.5 ft)......................................................................................................................................................... N
3 m (10 ft).......................................................................................................................................................... P

2) SPECIFY OUTPUT SIGNAL__________________________________________________________
3) SPECIFY RANGE LIMITS AND SPAN_________________________________________________
4) SPECIFY CALIBRATED RANGE______________________________________________________
5) SPECIFY SENSOR CODE___________________________________________________________
6) SPECIFY CONNECTING TUBING LENGTH___________________________________________
7) SPECIFY SENSOR DIMENSIONS____________________________________________________
A. Diameter (Y) [9.53 mm (0.375 in) standard]________________________________________
B. Length (X) [75 or 150 mm (3 or 6 in)—see Range Limits table]________________________
C. Extension length (J) [450 mm (18 in) standard]_____________________________________
D. Insertion length (U)______________________________________________________________
E. Immersion length (R) _____________________________________________________________
8) SPECIFY BUSHING EXTERNAL THREAD (if required)__________________________________
9) SPECIFY OPTIONAL FEATURES_____________________________________________________
10) SPECIFY INFORMATION FOR INSTRUMENT TAG AND APPLICATION________________

Page 30

Pneumatic Transmitters (Filter Regulators)
FRS Filter Regulators
Pressure Transmitters

FRS02 & FRS03

FRS02 Filter Regulatorn Filter regulator for all pneumatic instruments
n Aluminum housing FRS02
n Max. input pressure 10 bar
n Control range 0 to 8 barn Max. air flow 18 Nm³/hn Temperature range –20 ... +60 °C
n Gauge scale in bar / psi / kPan Sinter bronce filter 30 μm
n Explosion protection: ATEX II 2 GD c IIC T6
n Pneumatic Connection ¼’’ NPT
n Mounting bracket as accessories

Control of instrument supply airpressure, and removal of dust particles
and water content by filtration.
FRS03 Filter Regulator Stainless Steel 316
n Filter regulator for all pneumatic instruments
n Stainless Steel Housing 316
n Max. input pressure 10 bar
n Control range 0 to 8 barn Max. air flow 18 Nm³/hn Temperature range –40°C... +80°C
n Gauge scale in bar / psi / kPan Sinter SST filter 30 μm
n Explosion protection: ATEX II 2 GD c IIC T6
n Pneumatic Connection ¼’’ NPT
n Mounting bracket as accessories

Control of instrument supply airpressure, and removal of dust particles
and water content by filtration.

How to Order
Aluminum version Filter Regulator........................................................................................................ FRS02
Stainless Steel 316 version Filter Regulator.......................................................................................... FRS03
Mounting Bracket for FRS02 or FRS03................................................................................................ EBZG-FR1
Adjustable Mounting Bracket for FRS02 or FRS03............................................................................ EBZG-FR2

Page 31

Pneumatic Transmitters (Miscellaneous)
BYPASS MANIFOLDS, AIRSETS, 17BT, 17BS,45P

Bypass Manifolds

A variety of 3 and 5 valve manifolds is available
for fast, convenient zero checks and adjustmentof transmitters, and to isolate the transmitterfrom the process for maintenance.

17BT, 17BS Series Pneumatic
Buoyancy Transmitters for
Liquid Level Measurement

17BT and 17BS Series Pneumatic
Buoyancy Transmitters measure
and transmit liquid level interface
level, or density by sensingchanges in buoyant force exerted
on a fixed displacer element. They
can be flange-mounted to the top
or side of the vessel which may bevented, pressurized, or evacuated.
Displacer lengths range from
356 to 3810 mm (14 to 150 in) or
longer when required.

Page 32

Air Supply Pressure Regulators
and Filters (Airsets)

A broad selection of air pressure
regulators, filters, and combination
filter/regulators is available.
They are used with pneumatic
instruments requiring a regulated
filtered air supply.

45P Series Pneumatic
Indicating Transmitters

45P Series Pneumatic
Indicating Deflection TypeTransmitters measurepressure, temperature,differential pressure, or
liquid level, depending
on element selection, andtransmit a proportional
20 to 100 kPa or 3 to 15
psi signal.

Primary Measurement Elements

Primary Measurement Elements
Using primary measurement elements, you can adapt many instruments tospecific applications.
Foxboro makes elements for all types of measurements — flow, pressure,temperature, moisture, and humidity.

Orifice Plates and Flange Unions
Used with differential pressure meters to measure flow rate.

Pressure Seals
Ideal for measuring corrosive, viscous, or hot fluids.

Thermocouple/RTDs
Numerous configurations for use with transmitters,recorders, and controllers.

Thermowells
Protect thermocouples and RTDs from damaging process materials.

Dewcel Elements
The most accurate, reliable way to measure moisture and humidity.
These elements are designed for use with both pnuematic and electronic devices.
With them, you can count on the high performance and dependability you expectfrom all Foxboro instruments.

Page 33

Primary Measurement Elements (Orifice Plates, Flange Unions)Orifice Plates andFlange Unions

ORIFICE PLATES, FLANGE UNIONSn Proven Performance• Widely recognized as the industry 		
standard

n Versatile• Works with either a d/p Cell Transmitteror a bellows metern Broad Choice of Sizes and Materials• For just about any line size or application

Orifice Plates are used with differential
pressure meters to measure flow rate.

For complete specifications, refer to ProductSpecification Sheet PSS 3-5A1 A.

Orifice Plate Specifications
Material:
These plates conform to National Association ofCorrosion Engineer’s (NACE) Standard MR-01-75.
Flange Type Rating:
ANSI Class 150, 300, 400, or 600, as specified.
Bore:
Concentric, with sharp and square inlet edge. Plate
is beveled if required by conformity specifications.
Nominal Plate Thickness:
4 mm (0.125 in) for 25 to 200 mm (1 to 8 in) pipe
sizes, 7 mm (0.250 in) for 250 to 350 mm (10 to
14 in) pipe sizes, and 10 mm (0.375 in) for 400 to
500 mm (16 to 20 in) pipe sizes.

Legend:Upstream side of tab handle marked “inlet” andwith bore size and flange type/rating. Downstreamside marked with trademark, material, and tag.
Conformity (Specifications and Computations):
Finish: Number 4 (45 microinch or better).
Dimensions:
Meet the intent of applicable portions of ASME
“Fluid Meters” (Sixth Edition) and/or AGA Report
3 (1969), adopted as ANSI/API 2530.
Computations:
According to R.W. Miller’s “Flow Measurement
Engineering Handbook.” This handbook is available
from Foxboro. Refer to Part No. B0150YW.

Orifice Flange Union Specifications
Orifice Flange Unions are furnished in accordancewith latest AGA and ASME code recommendations.
Construction:
Unions include 2 flanges, 2 jackbolts with nuts, 2
composition gaskets, 4 pipe plugs, and bolting.
Material:
Flanges: ASTM A-105 carbon steel, ASTM A-182
F304 ss, or ASTM A-182 F316 ss, as specified.

Pressure Tap Connections:
2 sets per union, 180 degrees apart. Class
300 and 600 unions are threaded for 1/2-inch
pipe nipples. Class 900 and 1500 unions are
threaded for 3/4-inch pipe nipples. The pressure
tap hole diameter is: 6.35 mm (0.25 in) for 65
mm (2 1/2 in) pipe size and smaller, 9.53 mm(0.375 in) for 80 mm (3 in), and 12.7 mm (0.50
in) for 100 mm (4 in) pipe size and larger.

Bolting:With steel on stainless steel flanges;
ASTM A193 Grade B7 alloy steel stud bolts withASTM A194 Grade 2H nuts.

Optional Features
Reverse Flow:
For reverse flow applications, the plate is bored andfinished equally on both faces and the bore is not beveled. The plate must be used with symmetrical taps.
Preparation for Nuclear, Oxygen, or Chlorine
Service is Available.

Page 34

Optional Orifice Bores:
Eccentric, segmental, or quadrant edge bore.
Materials of Construction:
Hastelloy C, Monel, Alloy 20, and titanium areavailable.

Primary Measurement Elements (Orifice Plates, Flange Unions)
ORIFICE PLATES, FLANGE UNIONS

Optional Features (continued)
High Temperature and High Differential
Pressure Operation:
The Orifice Plate may warp if the temperature of
the process fluid exceeds 200°C (400°F) and thedifferential pressure (ΔP) is greater than 50 kPa(200 inH2O). Refer to Foxboro for determinationof the plate thickness for these conditions.

Drain or Vent Hole:
Plate drilled with recommended maximum
diameter hole to applicable standard.
Other materials of construction, flange
ratings, plate thickness, and line size areavailable.

How to Order
1) SPECIFY ORIFICE PLATE CODE:				____________SERIES 120 - _____ – _____

Plate, Foxboro Tab Type................................................................................ SERIES 120
Material: ASTM A-240 Type304 ss..............................................................................................................................................304
316 ss..............................................................................................................................................316

For use with Flange Type Rating: ANSI Class

150................................................................................................................................................................ 150
300................................................................................................................................................................ 300
400................................................................................................................................................................ 400
600................................................................................................................................................................ 600

2) SPECIFY NOMINAL LINE (PIPE) SIZE AND SCHEDULE NUMBER_______________________
3) SPECIFY ORIFICE DIAMETER OR SUPPLY COMPLETED FLOW DATA SHEET(Refer to Foxboro)_________________________________________________________________
4) SPECIFY OPTIONAL FEATURES_____________________________________________________
5) SPECIFY INFORMATION FOR IDENTIFICATION TAG_________________________________
1) SPECIFY FLANGE UNION CODE:_____ ____ - _____ – _____

Style:

Slip-On................................................................................................................. FU-S
Threaded............................................................................................................. FU-T
Welding-Neck.....................................................................................................FU-W

Flange Facing:

Raised Face........................................................................................................................ RF
Ring Joint (Not available with FU-T)............................................................................... RJ

Material:

ASTM A-105 Carbon steel........................................................................................................... CS
ASTM A-182 F304 ss.....................................................................................................................304
ASTM A-182 F316 ss.....................................................................................................................316

ANSI Rating:Class 300...................................................................................................................................................... 300
Class 600 (Not available with FU-T).......................................................................................................... 600
Class 900 (Not available with FU-T).......................................................................................................... 900
Class 1500 (Not available with FU-T)....................................................................................................... 1500

2) SPECIFY NOMINAL LINE (PIPE) SIZE________________________________________________
3)	SPECIFY PIPE SCHEDULE NUMBER OR ACTUAL INTERNAL DIAMETER FOR
WELDING-NECK TYPE UNIONS____________________________________________________
4) SPECIFY OPTIONAL FEATURES_____________________________________________________
5) FOR DIFFERENTIAL PRESSURE TRANSMITTER, REFER TO: Page 3.

Page 35

Primary Measurement Elements (Integral Flow Orifice Assemblies-IFOA)
Integral Flow Orifice
Assemblies (IFOA) Series

IFOA

n The IFOA has very high accuracy whenequipped with associated piping

n The IFOA can be used with any

differential pressure transmitter
having standard process connectionsn Process wetted materials are available

for use with both corrosive andnoncorrosive fluids

n The transmitter can be either inte­grally

coupled or remotely connected

n Process wetted material meets NACE

Standard MR-01-75

The IFOA Series Integral Flow Orifice
Assemblies adapt electronic and/or
pneumatic d/p Cell Transmitters formeasuring small flow rates.

For complete specifications, refer to
Product Specification Sheet PSS 3-5A1 B.

Specifications
Assemblies with Associated Piping:

Flange Material:
Forged 316 ss ASTM A182-F316.

Body Material:
Cast AISI Type 316 ss ASTM A351 Grade CF-8M
stainless steel.

Assemblies without Associated Piping:
Body Material:
Cast AISI Type 316 ss ASTM A351 Grade CF-8M
stainless steel or cast Hastelloy C Grade CW-2M
per ASTM A494/A494M-86, as specified.

Piping Material:
Seamless stainless alloy steel pipe to ASTM
A-312 Grade TP-316, Schedule 40 for the 15
and 25 mm (1/2 and 1 in) sizes, Schedule 80 for
the 40 mm (1 1/2 in) size.

Both Types:
Static Pressure Rating and Process Temperature Limits:
		
Assembly			
Static Pressure Rating and		
Size			Process Temperature Limits (a)End		Connectionmm		
in
SI Units		

U.S.Units		15		1/2
					

20
16

MPa from -40 to +40°C
MPa at 150°C

3000 psi from -40 to +100°F
2300 psi at 300°F

WELD
25		 1ENDS				

1410.5MPa from -40 to +40°C
MPa at 150°C

2000 psi from -40 to +100°F
1550 psi at 300°F

					
40		
1 1/2
		

5
4

MPa from -40 to +40°C
MPa at 150°C

750 psi from -40 to +100°F
580 psi at 300°F

15 & 25		 1/2 & 1
THREADED
ENDS
40		
1 1/2
					

10

MPa from -40 to +150°C

1500 psi from -40 to +300°F

54

MPa from -40 to +40°C
MPa at 150°C

750 psi from -40 to +100°F
580 psi at 300°F

FLANGED ENDS (b)		 All sizes				

Static Pressure Rating of Flange Selecteda) Process temperature limits are -40 and +150°C (-40 and +300°F). Transmitter temperature limitations must be observed
when integrally mounted to IFOA assembly. For higher pressures and temperatures, tb) Available with associated piping only.

Page 36

Primary Measurement Elements (Integral Flow Orifice Assemblies-IFOA)IFOASpecifications (continued)
Body Bolting:
ASTM A193 Grade B7 cadmium- or zinc-platedwith yellow chromate finish alloy steel stud bolts
and ASTM A194 Grade 2H nuts.

Orifice Plate Material:
316 ss ASTM A240, Monel ASTM B127, orHastelloy C276 ASTM B626, as specified.
Orifice Sealing Gasket Material:
Glass-reinforced ptfe.

Optional Features
Process wetted materials in compliancewith NACE Standard MR-01-75.

Orifice Kits:

High pressure assembly (IFO). Staticpressure rating 40 MPa (6000 psi).
17-4 PH ss body bolting.
Oxygen service preparation.
3-valve manifolds.
180° U-Bend Integral Orifice Attachment.

		
		
		

Consisting of 7 bored diameters perstandard specifications for 15 mm(1/2 in) assembly.

		
316 ss		Consisting of 5 bored diameters perstandard specifications for 25 mm(1 in) assembly.

		
		
		

Consisting of 5 bored diameters perstandard specifications for 40 mm(1 1/2in) assembly.

		
Monel
		

Consisting of 7 bored diameters perstandard specifications for 15 mm(1/2 in) assembly.

How to Order
1) SPECIFY MODEL NUMBER: IFOA	­– ___ __

__ __

__

__ – ___

Nominal Size:

15 mm (1/2 in).......................................................................................... 0H
25 mm (1 in).............................................................................................. 01
40 mm (1 1/2 in)....................................................................................... 1H

Body Material:

316 ss ................................................................................................................... S
Hastelloy C, Grade CW-2M (without associated piping only)...................... H

End Connection without Associated Piping:Socket Weld Body........................................................................................................1
Threaded Body (NPT)..................................................................................................2
Threaded Body (R metric)...........................................................................................8

End Connection with Associated Piping:

Pipe Ends Prepared for Welding....................................................................................... 3
Pipe Ends Threaded (NPT).................................................................................................. 4
Pipe Ends Threaded (R metric).......................................................................................... 9
Pipe Ends Flanged ANSI Class 150................................................................................... 5
Pipe Ends Flanged ANSI Class 300................................................................................... 6
Pipe Ends Flanged ANSI Class 600, 15 and 25 mm (1/2 and 1 in) only...................... 7

Orifice Plate Material:

316 ss .............................................................................................................................................SMonel ............................................................................................................................................ M
Hastelloy C276............................................................................................................................. H

HOW TO ORDER Continued on Next Page.

Page 37

Primary Measurement Elements (Integral Flow Orifice Assemblies-IFOA)IFOAHow to Order (continued)
1) SPECIFY MODEL NUMBER (continued): IFOA	­– ___ __

__ __

__

__ – ___

Orifice Bore Diameter: Nominal Size 15 mm (1/2 in):

0.508 mm (0.020 in).............................................................................................................................. A
0.889 mm (0.035 in).............................................................................................................................. B
1.524 mm (0.060 in).............................................................................................................................. C2.540 mm (0.100 in)..............................................................................................................................D
4.064 mm (0.160 in).............................................................................................................................. E
6.350 mm (0.250 in).............................................................................................................................. F
8.890 mm (0.350 in)..............................................................................................................................G
Jewel orifice. 316 ss base material. 0.0508 to 0.381 mm (0.002 to 0.015 in).............................. U
Nonstandard within beta limits of 0.1 to 0.8..................................................................................... V

Orifice Bore Diameter: Nominal Size 25 mm (1 in):6.147 mm (0.242 in).............................................................................................................................. H
8.661 mm (0.341 in).............................................................................................................................. J12.14 mm (0.478 in).............................................................................................................................. K
16.64 mm (0.655 in).............................................................................................................................. L
21.13 mm (0.832 in)..............................................................................................................................M
Nonstandard within beta limits of 0.1 to 0.8..................................................................................... V

Orifice Bore Diameter: Nominal Size 40 mm (1 1/2 in):

9.703 mm (0.382 in)..............................................................................................................................N
13.67 mm (0.538 in).............................................................................................................................. P
19.08 mm (0.751 in).............................................................................................................................. R
25.86 mm (1.018 in).............................................................................................................................. S
29.97 mm (1.180 in).............................................................................................................................. T
Nonstandard within beta limits of 0.1 to 0.8..................................................................................... V

Optional Connectors: For remote mounting d/p Cell Transmitter:

316ss, 1/2NPT................................................................................................................................................. 1S
316 ss, R1/2..................................................................................................................................................... 2S
Hastelloy C, 1/2 NPT...................................................................................................................................... 1H
Hastelloy C, R1/2............................................................................................................................................ 2H2) SPECIFY ORIFICE BORE DIAMETER OR SUPPLY COMPLETED FLOW DATA SHEETS(Refer to Foxboro) FOR NONSTANDARD OR JEWEL ORIFICE__________________________
3) SPECIFY OPTIONAL FEATURES______________________________________________________
4) SPECIFY INFORMATION FOR INSTRUMENT TAG_____________________________________

Page 38

Primary Measurement Elements (Diaphragm Pressure Seals)
DIAPHRAGM PRESSURE SEALS

Diaphragm Pressure Seals
n Foxboro offers a comprehensive line of

Diaphragm Pressure Seals. These seals
are used to separate pressure
instruments from process fluids.

n Separation is required if the process:• is corrosive• is viscous enough to clog
• contains suspended solids• could freeze or solidify• is toxic• requires sanitary handlingn Diaphragm Pressure Seals areavailable in a wide range of designs andmaterials

For complete specifications, refer to
Product Specification Sheet PSS 3-2C1 A.

Diaphragm Pressure Seals are forconnection to pressure instruments—
to isolate the process from the
instrument and piping.

Diaphragm Pressure Seals
Selection 			Description				Process Connection:
			 PES-SB: 1/4,1/2, 3/4,1,1 1/4, or			 1 1/2 NPT
THREAD		 PES-SG: 1/4,1/2,3/4, or 1 NPTATTACHED		
Operating Pressure:					 Maximum working pressure:			 0 to 17.5 MPa (0 to 2500 psi).

Illustrations and AS References (a)As Illustrated..........................AS Reference PES-SB
With 1/4-inch flushingconnection............................. AS Reference PES-SG		
Process Connection:
			 ANSI raised face flange.
FLANGE
Operating Pressure:
ATTACHED		 Maximum working pressure:			 Governed by flange rating.
					
				
					

As Illustrated......................... AS Reference PES-SC
With 1/4-inch flushingconnection............................. AS Reference PES-SD		
Process Connection: Welded to process line.
		
IN-LINE				
FLOW­Operating Pressures:		
THROUGH		
Maximum working pressure: Governed by user’s weld,
			 0 to 10.5 MPa at 38 C (0 to 1500 psi at 100’F).

80 mm (3 in) line only.100 mm (4 in) line & larger
As Illustrated.......................... AS Reference PES-SJ					As Illustrated.........................AS Reference PES-SMa) Other types available. Wetted parts materials available: Steel, AISI Types 304 and 316 ss, Carpenter 20, Monel, nickel, ....
titanium, Hastelloy B and C, PVC,polypropylene, ptfe, and Inconel.

Page 39

Primary Measurement Elements (Thermowells and DEWCEL)THERMOWELLS AND DEWCEL

Thermowells
n Rigid material, concentricity specifications, andindividual hydrostatic testing assures maximum
soundness and highest qualityn Connections:• Flanged (ANSI and ISO): Variety of sizes and 		pressure ratings• Threaded (metric and NPT): R1/2, R3/4, and 		R1 (1 /2, 3/4, and 1 NPT)n Type:• Drilled construction, welded construction, 		plain, with lagging, sanitary, straight, tapered, 		
restricted tip

Foxboro Thermowells separate thetemper­ature measuring sensitive portionof an RTD, Thermocouple, or Filled
Thermal System from a potentiallycorrosive or damaging process media.
These wells permit ready removal of thesensor without process shutdown.

n Materials:
Brass
Carpenter 20 Cb-3
C1018 Carbon Steel
R-Monel 405
304 ss (standard)
304 Low Carbon(304L) ss
316 ss (standard)

347 ssF11 Alloy SteelF22 Alloy Steel
Hastelloy B
Hastelloy C 276
Inconel 600
Nickel 200
Titanium

316L ss
Aluminum
CopperK-Monel309 ss
310 ss
410 ss
446 ssDew Point Measurement
System — DEWCELn The DEWCEL System provides a direct readingof absolute humidity in ambient temperature
between -45 and +105°C (-50 and +220°F)

When connected to a temperature measurement device, the DEWCEL System measuresabsolute humidity. By means of appropriate
tables, the temperature can be converted to
dew point, percent water vapor, parts per million, or other units of absolute moisture. In practice, the conversion is accomplished in recorderchart layouts, characterized amplifiers, or computer programs.
n Accuracy: ±0.8°C (±1.5°F) at 32°C (90°F)

The DEWCEL is used with RTD, thermo­
couple, or filled thermal systemtemperature measuring equipment.

dew point (not including temperature sensor
tolerances)n Dew Point Range Limits: -45 and +60°C (-50 and+140°F)n Relative Humidity Limits: 12 and 100% relative

humidityn Pressure Rating: Zero Pa absolute to 865 kPa

gauge (zero psia to 125 psig)

Page 40

Large Case Instruments

Large Case Instruments
Foxboro large case instruments have been on the job for years in process plants
around the world.
These versatile instruments fit a wide range of applications. There’s a unit for justabout every process function:• Recording• Indicating• Controlling• Totalizing• Transmitting
And they are available with a selection of measurement elements:• Flow• Pressure• Humidity• Temperature• Servo Elements for RTDs/Thermocouples and other Electrical Inputs
You can also choose from an extensive list of options such as electrical contacts,
integrators, batch controls, pen types, control modes—and numerous other features.
Foxboro large case instruments can be field mounted on pipe stands, in panels, oron flush surfaces. And their durable housings are resistant to moisture, corrosion,and damaging environments.
Their dependability is legendary. That’s why process engineers specify them timeand again.

Page 41

Large Case Instruments (Controllers)
40P, 40M Series Recording
and Indicating PneumaticControllers

40P, 40M

n Precise Control
n Chart Records or Indicationn Weatherproof Construction (40P):

NEMA Type 3

n Wide Selection of Measuring Elements
n Versatile Mounting: Surface, Panel, or 		

Pipe Mounting

The 40P, 40M Series Recording and Indicating
Pneumatic Controllers receive a mechanicalinput from an integrally mounted measuring
element, compare it to a set point, andproduce a pneumatic output signal.

For complete specifications, refer to
Product Specification Sheets PSS 3-1A1 A
and 3-1A2 A, B, C, and F.

Functional Specifications
Single Action:
One control unit per 40P or 40M case. Dual,
duplex, and auto-selector versions are alsoavailable.

Control Modes:
See “HOW TO ORDER.”
Measurement Elements:
See “HOW TO ORDER.”
Controller Action:
The output signal either increases or decreases
with increasing measurement, as specified.

Recorder Chart Drives:
Electrical: The standard speed is 1 revolution
per 24 hours for nominal 120 or 240 V, 50 or 60
Hz, as specified.
Mechanical: The standard speed is 1 revolution
per 24 hours with a 24-hour movement.

Output Signal:
See “HOW TO ORDER.”
Supply Pressure:
140 kPa (20 psi, 1.4 bar or kg/cm2).

Battery Operated: 11 Speed
Recorder Pens:
Fiber tip disposable or box.

Physical Specifications
Enclosure:
40P Series: The case and door are glass-fiber
reinforced polyester moldings, compounded
for superior corrosion resistance. The overall
construction provides the environmental
protection of IEC IP53 and NEMA Type 3.
40M Series:
Durable, die-cast aluminum case and door withcorrosion-resistant vinyl finish.

Mounting:
Standard mounting is flush in a panel up to 16
mm (0.6 in) thick or on a surface. A kit of partsfor vertical mounting on a DN 50 or 2 in pipe isavailable (40P Series only).
Expendable Accessories:
Charts: 100 HUMITEX nominal 300 mm (12 in)circular charts [nominal 100 mm (4 in) calibrated
scale] are supplied with each instrument.
Ink:
1 U.S. fl oz (30 cm3) is supplied for each box
type pen.

Page 42

Large Case Instruments (Controllers)
40P, 40M

Performance Specifications
Accuracy:
See Measurement Element selected.

Repeatability:
0.25% of calibrated span.

Optional Features
Pneumatic Set:
For remote set point adjustment.

Batch:For batch control.

External Manual Set Point Knob.

Type 70 Contacts:
For electrical contact output. Refer to Page 57.

Ratio:
Maintains ratio between uncontrolled andcontrolled variable.

External Connection to Integral Bellows.

How to Order1) SPECIFY MODEL NUMBER:						A
40P or 40M	­___ -	­___	­___	­___ -	­___	­___	­___	­___	­___ /	­____________
Function:
Recording............................................................R
Indicating............................................................ N

Action:	Single (One control unit actuated by one
measuring element. Control will apply to first
listed element.)......................................................... A

Control:

On/Off...................................................................................1
Full proportional 0.5 to 200%...........................................2
Full proportional plus derivative 0.01 to 50 minutes.....3
	Full proportional plus integral (reset) 0.01 to
50 minutes per repeat...................................................4
Full proportional plus integral (reset) plus derivative....5
Narrow band proportional 0.5 to 25% or 1.0 to 50% ...6

Automatic/Manual Internal Transfer Switching:

Without internal regulator, without transfer switch (a)........... J
2-position nozzle seal switch (b)............................................... L
Balanceable automatic/manual unit - right (b)........................N
Balanceable automatic/manual unit - left (b).......................... P

Output Signal and Gauge:

20 to 100 kPa signal, 150 kPa gauge................................................5
3 to 15 psi signal, 22 psi gauge.........................................................2
0.2 to 1.0 bar signal, 1.5 bar gauge..................................................6
0.2 to 1.0 kg/cm2 signal, 1.5 kg/cm2 gauge.....................................4

Mounting:

Flush (c)..........................................................................................................F
Pipe (c), (d).....................................................................................................P
Surface (c).......................................................................................................S
Yoke. 40P Series available with Type 37 Element only...........................Y(a) Available with Control code 1 only. (b) Not available with Control code 1. (c) Not available with 40P Series with Type 37 Element.
(d) Not available with 40M Series.
HOW TO ORDER Continued on Next Page.

Page 43

Large Case Instruments (Controllers)
40P, 40M

How to Order (continued)1) SPECIFY MODEL NUMBER:						A
40P or 40M	­___ -	­___	­___	­___ -	­___	­___	­___	­___	­___ /	­____________
Chart Drive or Scale:Electrical - 24 h rotation, 120 V, 60 Hz (options available).............................. E
Mechanical - 24 h rotation and wind (options available)................................M
Sector .....................................................................................................................N

Pens or Pointers:

1 pen or pointer............................................................................................................. 1
2 pens or pointers.......................................................................................................... 2
3 pens .............................................................................................................................. 3

Pen Type or Scale Type:Fiber tip disposable................................................................................................................F
Box .......................................................................................................................................B
Single range scale...................................................................................................................S
Double range scale................................................................................................................ D

Elements: (Refer to Pages 48-56 for Element Descriptions)Type 37 (a)..........................................................................................................................................DE-A, B
Absolute pressure bellows................................................................................................................PA-CA
Receiver................................................................................................................................................... PC
Absolute pressure double spiral...................................................................................................... PA-MA
Pressure helical.............................................................................................................................. PB-M, PB-AA
Pressure spiral..................................................................................................................................... PB-BA
Pressure bellows............................................................................................................................. PB-CA, CC
Diaphragm........................................................................................................................................PB-DF, PF
Stainless long helical.......................................................................................................................... PB­GA
Temperature.................................................................................................................................... TA-1A, 2A,													 2B, 3B
Electronic servo....................................................................................................................................... E(a) Only one per instrument.

2) SPECIFY MEASUREMENT RANGE___________________________________________________
3) SPECIFY CONTROLLER ACTION____________________________________________________
4) SPECIFY OPTIONAL FEATURES_____________________________________________________
5) SPECIFY INFORMATION FOR INSTRUMENT TAG____________________________________

Page 44

Large Case Instruments (Controllers)
43AP Series Pneumatic
Indicating Controllers

43AP

n Wide Selection of Direct Connected

Measurement Elements

n Broad Choice of Control Modes• On/off, proportional, proportional plus 		
derivative, proportional plus integral (reset),
proportional plus integral (reset) plus deriva		tive, differential gap, and automatic shutdown
n Internal Bumpless Auto/Manual Transfer• Smooth internal switchingn Weatherproof• Glass-fiber reinforced case, gasketed door
• NEMA Type 3 protectionn Accuracy Unaffected by Mounting

The 43AP Series Pneumatic
Indicating Controllers continuously
indicate and control pressure,vacuum, temperature, or flow.

• Measurement element and control unit
mounted on rigid steel plate

For complete specifications, refer to
Product Specification Sheet PSS 3-1B3 A.

Functional Specifications
Controller Action:
Output signal either increases or decreases withincreasing measurement, as specified; action isreversible in the field.

Control Modes:
See “HOW TO ORDER.”
Measurement Elements:
See “HOW TO ORDER.”
Output Gauge:
0 to 200 kPa, 0 to 30 psi, or 0 to 2 bar or kg/cm2,
as specified.
Set Point Adjustment:
By means of a knob mounted inside the case.

Supply Pressure:
140 kPa, 20 psi, 1.4 bar or kg/cm2.
Output Signal:
See “HOW TO ORDER.”
Ambient Temperature Limits:-40 and +80°C (-40 and +180°F).

Performance Specifications
Accuracy (When used with the elements aslisted under “HOW TO ORDER”): ±0.5% ofcalibrated span.

Repeatability:
0.2% of calibrated span.

Physical Specifications
Mounting:
Panel, surface, pipe, or yoke. Panel: Flush in apanel up to 16 mm (0.6 in) thick.
Surface: suitable for all controllers having
internally mounted elements. This mounting is notavailable with heavy duty helical elements. (Theseelements extend through the back of the case.)

Pipe: a kit of parts to fit a DN 50 or 2 in vertical pipe.
Yoke: a kit of parts to fit a vertical DN 50 or 2 inpipe stub. This mounting scheme is designed
specifically for controllers having rear-mounted
differential pressure elements.

Page 45

Large Case Instruments (Controllers)
43AP

Physical Specifications (continued)
Enclosure:
The case and door are glass-fiber reinforced
polyester moldings, compounded forsuperior corrosion resistance. The door
has a shatterproof polycarbonate window,
ultraviolet-resistant. The overall construction is
weatherproof, meets IEC IP53, and provides theenvironmental protection of NEMA Type 3.

Scale:
Black markings on a white background, sectorshaped with nominal effective length of 150 mm(6 in). Refer to Chart and Dial Catalog 600 foravailable ranges.

Optional Features
Integral Air Supply Set.
Type 70 Contacts:
For electrical contact output. Refer to Page 57.

Batch:For batch control.
Vent Connection.

Control Valve Mounting.

External Phenolic Nameplate.

External Connection to Integral (Reset) Bellows(ECRB) .

Glass Window in Door.

External Set Point.

Stainless Steel Data Plate.
Tamperproof Door Knob.

How to Order
1) SPECIFY MODEL NUMBER: 43AP-___ ___ ___ ___ - ___ / ______________­

Mounting:

Field (pipe or yoke).............................................................................. F
Panel or surface.................................................................................... P

Control:

On/Off........................................................................................................... A1
Proportional 4 to 400%.............................................................................. A2
Proportional plus derivative 0.05 to 50 minutes..................................... A3
Proportional plus integral (reset) 0.01 to 50 minutes per repeat......... A4
Proportional plus integral (reset) plus derivative................................... A5
Differential gap 1 to 100%......................................................................... A7
Automatic shutdown................................................................................... A8

Output Signal and Gauge:

20 to 100 kPa signal, 200 kPa gauge................................................................. 5
3 to 15 psi signal, 30 psi gauge.......................................................................... 2
0.2 to 1.0 bar signal, 2 bar gauge...................................................................... 6
0.2 to 1.0 kg/cm’ signal, 2 kg/cm2 gauge......................................................... 4

Automatic/Manual Internal Transfer Switching:None ............................................................................................................................. N
Bumpless with 2-position switch, balance gauge, regulator.................................. C
2-position nozzle seal switch for manual control...................................................... D

Optional Suffix:

Remote pneumatic set point (not available with pressure element codes
		 PA-CA, PA-CC, PB-CA, PB-GA, PB-PF; servo element code E;
		 Type 70 Contacts; or options ECRB and “BATCH”)..................................................... P
HOW TO ORDER Continued on Next Page.

Page 46

Large Case Instruments (Controllers)
43AP

How to Order (continued)
1) SPECIFY MODEL NUMBER: 43AP-___ ___ ___ ___ - ___ / ______________­

Elements: (Refer to Pages 48-56 for Element Descriptions)Type 37 (43AP-F only).......................................................................................................................DE-A, B
Absolute pressure bellows............................................................................................................ PA-CA, CC
Absolute pressure double spiral...................................................................................................... PA-MA
Pressure helical.................................................................................................................................... PB-AA
Pressure spiral..................................................................................................................................... PB-BA
Pressure bellows............................................................................................................................. PB-CA, CC
Diaphragm........................................................................................................................................PB-DF, PF
Stainless long helical (43AP-F only)..................................................................................................PB-GA
Receiver................................................................................................................................................... PC
Temperature.................................................................................................................................... TA-1A, 2A,									 2B, 3B
Electronic servo....................................................................................................................................... E

2) SPECIFY CONTROLLER ACTION____________________________________________________
3) SPECIFY OPTIONAL FEATURES_____________________________________________________
4) SPECIFY INFORMATION FOR INSTRUMENT TAG AND APPLICATION______________________________________________________________________________________________________

Page 47

Large Case Instruments (Controllers)
P417

P417 Pneumatic Field
Controller
n Same basic instrument for field / panel

mountingn Aluminium die-cast housing, with lock,protection class IP 54 (IP 65)

Complete instrument
P417pneumatic controllers are complete
pneumatic control stations and, in conjunction
with transmitters, are suitable for all control
applications in process plants, particularly forfield mounting. Field controllers are also available
without an indicator.

}

Measured value x............ 0.2 to 1.0 bar /Set point w....................... 3 to 15 psi /Output y............................ 20 to 100 kPa
Supply air.......................... 1.4 ± 0.1 bar / 20 ±1.4 psi /140 ± 10 kPa
Inherent air
consumption 1) 2)............. ≤ 500 l / hr
Housing
Front finish........................ door enamelled grey to
RAL 7000, with lock
Service label..................... 50 x 13mm, replaceable
Materials
Housing, with door......... aluminium, with DD enamel,grey RAL 7000
Connection block forwall / panel mounting.... aluminium, with DD enamel,grey RAL 7000
Measuring elements....... copper-beryllium
Other internal
components..................... aluminium, brass, galvanizedand passivated steel
Double indicator for x and w
Error limit.......................... < 1%
Indicator for output signal
Indication range.............. 0 to 1.2 bar / 18 psi / 120 kPa
Error limit.......................... < 1.6%
Signal generators for set point and output signal
Output adjustable........... 0 to 1.2 bar / 18 psi / 120 kPa
Supply air dependency..≤ 0.25 % / 0.1 bar
Air consumption.............. ≤ 100 l / hr per signal
generatorn Well-established cross-bellows control systemn Set point, measured value and output indicationn Versions for master and slave control
n Push-buttons for manual / auto transfer
n Bumpless manual / auto transfer
n Limiting of controller output
n External feedbackn Field controller changeover relayn Mode switchingn Operating point set internally / externallyn Function altered by interchanging standard

modulesn Service label

Manual-Auto switch
Type.....................................push-buttons
Controller section
System.................................cross-belows
Control modes...................P, PD, PI, PID
Operating point P / PD
controller.............................set pneumatically, either
internally or externally
Proportional band Xp.......5 to 400%and higher
Gain Kp................................20 to 0.25 and lower
Reset time Tn......................0.03 to 50min
Derivative time Tv..............0.03 to 20min
Tracking error1)..................≤ 0.7 %Alignment offset1).............≤ 0.7 %Supply air dependency1)....≤ 0.2 % / 0.1 bar
Temperature effect1)........≤ 0.2 % / 10 K
Sensitivity1)..........................≤ 0.1 %Startup relay.......................for bridging time function“manual”
Tracking relay.....................for bumplessmanual - autotransfer, even with deviation
Mode switching:— PI-switching.....................reset bellows loading input,
internal or external— P / PI-switching...............switchingwith external signal
External feedback.............external pressure applied tofeedback bellows R+
Additional equipment
Min- /Max-limiting.............formin. ormax. limiting of“R” or “yA + R”
limit value settable to
0.2 to 0.6 bar or
0.6 to 1.0 bar
1) Measured at Xp = 100 % resp. with PI controller
2) Master version

Page 48

Large Case Instruments (Controllers)
Code P417-541
PID Controller withindicator, Tn and Tv
startup relays

Panel mounting

How to Order
Pneumatic Field Controller
Version

P417

Master controller with signal generator for output signal......................................... -1
Slave controller with signal generator for output signal............................................ -2
Master controller without signal generator for output signal................................... -3
Slave controller without signal generator for output signal..................................... -4
Field controller with indicator (5 pipe).......................................................................... -5
Field controller without indicator (5 pipe).................................................................... -6

HOW TO ORDER Continued on Next Page.

Page 49

Large Case Instruments (Controllers)
P417

How to Order (continued)
Control mode and equipment
P Operating point internally........................................................................................................................1
P Operating point externally (a)................................................................................................................12
PD with Tv-startup relay O.P. internally (b)..............................................................................................21
PD with Tv-startup relay O.P. externally (b).............................................................................................22
PI with Tn-startup relay (d)...........................................................................................................................31
PI with Tn-startup relay and tracking relay (d) ........................................................................................32
PI with PI-switch, YF internally (a)................................................................................................................3
PI with PI-switch, YF externally (b)..............................................................................................................34
PI with PI-switch,Tn-startup relay, YF internally (b).................................................................................35
PI with PI-switch,Tn-startup relay, YF externally (b)................................................................................36
PI externally feedback with Tn-startup relay (b)......................................................................................37
PI Auto-Selector-Master controller with Tn-startup relay (b)................................................................38
PI Auto-Selector-Slave controlle with Tn-startup relayr (b)...................................................................39
PID with Tn-, Tv-startup relay (d).................................................................................................................41
PID with Tn-, Tv-startup relay and tracking relay (d)...............................................................................42
PID with PI-switch, YF internally (b)............................................................................................................43
PID with PI-switch, YF externally (b)...........................................................................................................44
PID with PI-switch, Tn-startup relay, YF internally (b)..............................................................................45
PID with PI-switch, Tn-, Tv-startup relay, YF externally (b)......................................................................46
PID externally feedback with Tn-,Tv-startup relay (b).............................................................................47
PID Auto-Selector-Master controller with Tn-, Tv-startup relay (b)......................................................48
PI Auto-Selector-Slave controller with Tn-, Tv-startup relay (b)............................................................49
P / PI with Tn-startup relay, O.P. internally (b)...........................................................................................51
P / PI with Tn-startup relay, O.P. externally (b)..........................................................................................52
P / PI with Tn-startup relay, tracking relay Operating point internally (b)..............................................53
P / PI with Tn-startup relay, tracking relay Operating point externally (b)..............................................54
P / PI with remote switching,Tn-startup relay Operating point internally (b)........................................55
P / PI with remote switching,Tn-startup relay Operating point externally (b).......................................56
P / PI / PID with Tn-startup relay, O.P.internally (b)......................................................................................61
P / PI / PID with Tn-,Tv-startup relay, O.P.externally (b)...............................................................................62
P / PI / PID with Tn-startup relay,tracking relay Operating point internally (b).....................................63
P / PI / PID with Tn-startup relay,tracking relay Operating point externally (b).....................................64
Indicating Range (Scale) for X and W
0 - 100% linear (h)........................................................................................................................................................ A
to be specified (acc.standard list) (h)....................................................................................................................... B
without indicator (k)..................................................................................................................................................... C
Signal Range
0.2-1.0bar................................................................................................................................................................................. 1
3-15psi .................................................................................................................................................................................... 2
20-100kPa................................................................................................................................................................................. 3
Indicating Range for output signal
0-1.2bar ...............................................................................................................................................................................................1
0-18psi ...............................................................................................................................................................................................2
0-120kPa..............................................................................................................................................................................................3
Type of mounting
Wall / Pipe mounting, pneum.conn. 1/4 -18 NPT................................................................................................................................. A
Wall / Pipe mounting, pneum.conn. G 1/4 ............................................................................................................................................ B
Panel mounting, pneum.conn. 1/8 -27 NPT........................................................................................................................................... C
Optionsmin. limitation of YA and R (for P-controller YA only)......................................................................................................................................-A
min. limitationofR (b) (e)........................................................................................................................................................................................-B
max. limitation of YA and R (for P-controller YA only).....................................................................................................................................-C
max. limitationofR (b) (e).......................................................................................................................................................................................-D
min.- max. limitation of YA and R (for P-controller YA only) (b) (f).................................................................................................................-E
Kit for pipe mounting........................................................................................................................................................................................... -M
Plug for test jack (O.P. for P-, PD-controller)......................................................................................................................................................-S
Protection class IP 65 (for mounting type A, B)................................................................................................................................................-P
Tag.No. Labeling with inscription....................................................................................................................................................................... -L
Certificates
EN 10204-2.1 (DIN 50 049-2.1)..........................................................................................................................................................................-1
NOTES(a) Only available with Version -1, -2, -3, -4(b) Only available with Version -1, -2(c) Only available with Version -3, -4

Page 50(d) Only available witht Version -1,-2,-5,-6(e) Only available with control mode 31, 32, 41, 42(f) Only available with control mode 31, 32, 34, 36(g On request(h) Not available with Version -6(k) Not available with Version -1, -2, -3, -4, -5

Large Case Instruments (Recorders and Indicators)
40P, 40M Series Recorders
and Indicators

40P, 40M

n Wide Selection of Measuring Elements
n Versatile Mounting: Surface, Panel, orPipe Mounting

n Choice of Circular Chart Recorder

or either Concentric or Sector Scale
Indicatorn Weatherproof Construction (40P Series):

NEMA Type 3

For complete specifications, refer to
Product Specification Sheets PSS 3-1A1 A
and 3-1A2 A, B, C, and E.

The 40P and 40M Series Instruments
continuously record or indicate measurement of one or more process variables.

Functional Specifications
Measurement Elements:
See “HOW TO ORDER.”

Battery Operated: 11 speed.

Recorder Chart Drives:
Electrical: The standard speed is 1 revolution
per 24 hours for nominal 120 or 240 V, 50 or 60
Hz, as specified.

Ambient Temperature Limits:-30 and +60°C (-20 and +140°F).
Recorder Pens:
Fiber tip disposable or box.

Mechanical: The standard speed is 1 revolution
per 24 hours with a 24-hour movement.

Performance Specifications
See Measurement Element selected.

Physical Specifications
Enclosure:
40P Series: The case and door are glass-fiber
reinforced polyester moldings, compounded
for superior corrosion resistance. The door
has a shatterproof glass window. The overall
construction provides the environmental
protection of IEC IP53 and NEMA Type 3.
40M Series: Durable, die-cast aluminum case anddoor with corrosion-resistant epoxy powder finish.
Mounting:
Standard mounting is flush in a panel up to 16
mm (0.6 in) thick or on a wall. A kit of parts forvertical mounting on a DN 50 or 2 in pipe isavailable as an option (40P Series only).
Cardboard Nameplate:
A cardboard nameplate is supplied for
displaying the chart factor.

Expendable Accessories:
Charts: 100 HUMITEX nominal 300 mm (12 in)circular charts [nominal 100 mm (4 in) calibrated
scale] are supplied with each instrument.
Ink:
1 U.S. fl oz (30 cm3) is supplied for each box pen.
Indicator Pointer:
Sector indicators have an orange pointer andconcentric indicators have a black pointer.
Indicator Scales:
Black markings on white background. Refer toChart and Dial Catalog 600 for available ranges.
Sector:
The effective length is 171 mm (6.75 in).
Concentric:
The effective length is 594 mm (23.4 in).

Page 51

Large Case Instruments (Recorders and Indicators)
40P, 40M

Optional Features
Incandescent Type Internal Illumination.

Indicating Scale Transparent Mounted in
Instrument Door.

Bakelite Nameplate.

Plexiglass Circular Disc.

Flush Door Lock.

Portable Mounting.

Segmental Indicating Scale.

How to Order
1) SPECIFY MODEL NUMBER: 40P or 40M-___ - ___ ___ ___ ___ ___ / ______________­

Function:

Circular Chart Recorder..............................................................R
Sector Scale Indicator................................................................. N
Concentric Scale Indicator......................................................... KInternal Mechanism:

Recorder - standard............................................................................. R
Indicator- sector...................................................................................N
Indicator - concentric........................................................................... K
With Type 70 Contacts (recorder or indicator)................................ E

Mounting:

Flush (a)..........................................................................................................F
Pipe (a), (b).....................................................................................................P
Surface (a).......................................................................................................S
Yoke. 40P Series available with Type 37 Element only...........................Y

Chart Drive or Scale:Electrical, 24 h rotation, 120 V, 60 Hz (options available)............................... E
Mechanical, 24 h rotation and wind (options available)................................. M
Sector scale............................................................................................................N
Concentric scale.................................................................................................... K
Battery operated (11 speed)..............................................................................XX

Pens or Pointers:

1 pen (R) or 1 pointer (N or K)..................................................................................... 1
2 pens (R) or 2 pointers (N).......................................................................................... 2
3 pens (R)....................................................................................................................... 3
4 pens (R)....................................................................................................................... 4

Pen Type or Scale Type:Fiber tip disposable............................................................................................................... F
Box ...................................................................................................................................... B
Single range scale.................................................................................................................. S
Dual range scale. For 40PN and 40MN Series only.......................................................... D(a) Not available with 40P Series with Type 37 Element. (b) Not available with 40M Series.
HOW TO ORDER Continued on Next Page.

Page 52

Large Case Instruments (Recorders and Indicators)
40P, 40M

How to Order (continued)
1) SPECIFY MODEL NUMBER: 40P or 40M-___ ___ ___ ___ ___ - ___ / ______________­

Elements: (Refer to Pages 48-56 for Element Descriptions)Type 37 (yoke mounting only) (a)....................................................................................................DE-A, B
Absolute pressure bellows................................................................................................................PA-CA
Receiver................................................................................................................................................... PC
Absolute pressure double spiral...................................................................................................... PA-MA
Pressure helical.............................................................................................................................. PB-M, PB-AA
Pressure spiral..................................................................................................................................... PB-BA
Pressure bellows............................................................................................................................. PB-CA, CC
Diaphragm........................................................................................................................................PB-DF, PF
Stainless long helical..........................................................................................................................PB-GA
Temperature.................................................................................................................................... TA-1A, 2A,										 2B, 3B
Electronic servo....................................................................................................................................... E
Relative humidity (b)............................................................................................................................MA-K(a) Only one per instrument.
(b) Not available with 40P Series.

2) SPECIFY OPTIONAL FEATURES_____________________________________________________
3) SPECIFY INFORMATION FOR INSTRUMENT TAG____________________________________

Page 53

Large Case Instruments (Recorders and Controllers)
Oil & Gas Industry Flow
Recorders and Controllers

43APG

n 43APG Series Controllers:• NACE compliant element materials available.
• Low air consumption and internal corrosion 				resistance means the 43APG can be 				 operated directly from the process gas• Wide choice of measurement elements

n Glass-fiber reinforced case with gasketed

43APG Series Controllers provide precise
indication and control of flow, pressure,vacuum, and temperature.

door provides NEMA Type 3 protection.

For complete specifications, refer to ProductSpecification Sheets PSS 7-3C3 A and B.

Functional Specifications
Measurement Elements and Configurations:
See “HOW TO ORDER.”
Controller Output Signal and Supply Pressure:Output Signal

Supply Pressure

20 to 100 kPa (a)3 to 15 psi
0.2 to 1.0 bar or kg/cm2 (a)140 kPa (a)20 psi
1.4 bar or kg/cm2 (a)(a) Not applicable to the 43APG Series Controllers.

WARNING:EXPLOSION HAZARD CAN RESULT FROM USE
OF NATURAL GAS AS SUPPLY. INSTRUMENT
LOCATION MUST BE PROPERLY CLASSIFIED PER
NFPA-70. TOXICITY HAZARD CAN RESULT FROMUSE OF SUPPLY GAS CONTAINING HYDROGENSULFIDE (H2S). AREA MUST BE VENTILATED OR
BREATHING APPARATUS USED. REFER TO Ml
011-111.

Performance Specifications
Recorder Accuracy:
Spans less than 12:5 kPa, 50 inH20, 125 mbar ΔP
and all compound ranges: ±0.50% of span.

Repeatability: Less than 0.2% of span.

Controller (less element) Accuracy:

Hysteresis: ±1 % of output span.

Dead Band: Less than 0.1 % of span.

Physical Specifications
Door:
Hinged, glass-fiber reinforced phenylene oxide
molding with blue polyurethane finish. The clear
polycarbonate window is ultraviolet-resistantand shatterproof.

Case:
Glass-fiber reinforced, gray polyester molding.
Enclosure Classification:
Meets IEC IP53 and provides the environmental
protection of NEMA Type 3.

Optional Features
Refer to Foxboro.

Page 54

Large Case Instruments (Recorders and Controllers)
43APG

How to Order
1) SPECIFY MODEL NUMBER: 43APG-		 ___ ___ ___							2 ___ - ___ / ______________­
Mounting:

Field .................................................................................................... F
Panel or Surface................................................................................... P

Control:

On-Off........................................................................................................... A1
Proportional 4 to 400%.............................................................................. A2
Proportional plus Derivative 0.05 to 50 minutes.................................... A3
Proportional plus Reset (Integral) 0.01 to 50 minutes per repeat........ A4
Proportional plus Integral plus Derivative............................................... A5
Differential Gap 1 to 100%........................................................................ A7

Output Signal and Gauge:

3 to 15 psi signal, 30 psi gauge.......................................................................... 2

Automatic/Manual Internal Transfer Switching:

Bumpless with 2-position switch, balance gauge, regulator.................................. C
2-position nozzle seal switch for manual control...................................................... DNone ............................................................................................................................. N

Optional Suffix:

Remote Pneumatic Set Point (not available with element code
		 PA-CA, PA-CC, PB-PF, or PB-GA).................................................................................... P

Element: (Select one, refer to Pages 48-56 for Element Descriptions)Type 37 (43APG-F only)................................................................................................................. DE-A (a), B
Absolute pressure bellows............................................................................................................ PA-CA, CC
Receiver............................................................................................................................................... PC-OC
Absolute pressure double spiral...................................................................................................... PA-MA
Pressure helical.............................................................................................................................. PB-AA, AM (b)Pressure spiral............................................................................................................................... PB-BA, BM (b)Pressure bellows............................................................................................................................. PB-CA, CC
Diaphragm........................................................................................................................................PB-DF, PF
Stainless long helical (43APG-F only)...............................................................................................PB-GA
Temperature (c)............................................................................................................................... TA-1A, 2A,										 2B, 3B(a) Available in NACE compliant version.
(b) Element codes PB-AM and PB-BM comply with NACE Standard MR-01-75.
(c) Available with AISI Type 316 stainless steel tubing only.

2) SPECIFY SCALE FROM CHART AND DIAL CATALOG 600 OR REFER TO
FOXBORO________________________________________________________________________
3) SPECIFY CALIBRATED RANGE______________________________________________________
4) SPECIFY CONTROL ACTION (FIELD REVERSIBLE)____________________________________
5) SPECIFY INFORMATION FOR INSTRUMENT TAG____________________________________

Page 55

Large Case Instruments (Elements)
Type 37 Diaphragm
Differential Pressure Element

TYPE 37n I
deal for Liquid, Gas, and Steam Flow

Measurementn N
o Mercury Used: Safe Operationn A
mbient Temperature Compensationn Overrange Protectionn Requires No Power for Operation

For complete specifications, refer to
Product Specification Sheet PSS 3-4A2 A.
Type 37 Diaphragm Differential Pressure
Elements translate differential pressure toangular shaft position.

Functional Specifications
Spans:
Available between 5 and 50 kPa or 20 and 200
inH20.
Standard Differential Pressure Ranges:
0 to 5, 0 to 12.5, 0 to 25, and 0 to 50 kPa; or 0 to
20, 0 to 50, 0 to 100, and 0 to 200 inH20 ΔP.

Page 56

Maximum Process Pressure:
14 MPa (2000 psi).
Ambient Temperature Limits:-34 and +110°C (-30 and +230°F).

Large Case Instruments (Elements)
Pressure Measuring
Elements

PRESSURE MEASURING ELEMENTS

n W
 ide variety of element types,

materials, and pressure ranges

n Durable and dependable for longservice life

For complete specifications, refer to
Product Specification Sheet PSS 3-2A1 A.

Pressure Measuring Elements measure gauge
and absolute pressure and actuate recorders,
indicators, controllers, and transmitters.

Performance Specifications
Accuracy:
Better than ±0.5%.

Physical Specifications
Element Materials:
AISI Type 316 ss (316 ss).

Copper-Nickel-Manganese (Cu-Ni-Mn) Alloy.
K-Monel.

Bronze or Brass.

Optional Features
Overrange Protection, Underrange Protection:
Provides protection to the linkage and pointer,not the pressure element.

Oxygen Service Preparation:
Available for 316 ss spirals and 316 ss helicals.
Copper bearing brazing material will be incontact with process.

Page 57

Large Case Instruments (Elements)
PRESSURE MEASURING ELEMENTS

How to Order
1) SPECIFY ELEMENT CODE:___________­

					Normal Recorder Measurement Ranges
		Element			 (Torque Factor rating 30)
Type		Material
			

kPa		

psi,psia
12 psi15 psi24 psi............................... PC (a)			

80 kPa100 kPa—

Diaphragrn
Cu-Ni-Sn Pfinodal
75 mm (3 in)		

2 and 10 kPa
2 and 6 kPa vac

0.3 and 1.5 psi
0.3 and 0.9 Psi vac............................... PB-PF

Diaphragm
Cu-Ni-Sn Pfinodal
50 mm (2 in)		

6 and 70 kPa
6 and 35 kPa vac

0.9 and 10 psi
0.9 and 5 Psi vac............................... PB-DF

Receiver (Bellows)

Brass

35 and 200 kPa

4.5 and 29 Psi............................... PB-CA

Bellows

30 and 180 kPa
34 and 100 kPa vac

4 and 26 psi
4.9 and 15 psi vac...............................PB-CC		

82 and 1400 kPa
82 and 100 kPa vac

12 and 200 psi
12 and 15 psi vac............................... PB-BA		

316 ssBrass			
316 ssSpiral				

K-Monel

140 and 1400 kPa

20 and 200 psi

Helical		

316 ss
K-Monel

1400 and 40 000 kPa
1700 and 14 000 kPa

200 and 6000 psi
250 and 2000 psi

Heavy Duty Helical

316 ss

500 and 200 000 kPa

75 and 30 000 psi

Absolute Bellows
		

316 ssBronze17 and 240 kPa abs
13 and 270 kPa abs

2.5 and 35 psia
1.9 and 39 psia

Absolute Double Spiral316 ss140 and 700 kPa abs

20 and 100 psia...............................PB-BM............................... PB-AA...............................PB-AM...............................PB-GA...............................PA-CA...............................PA-CC.............................. PA-MA(a) Code PC-OC when used with the 43APG Series Controller on Page 46.

2) SPECIFY MEASUREMENT RANGE___________________________________________________
3) SPECIFY OPTIONAL FEATURES_____________________________________________________
4) SPECIFY INFORMATION FOR INSTRUMENT TAG____________________________________

Page 58

Large Case Instruments (Elements)
Filled Thermal System
Elements

THERMAL ELEMENTSn D
ependable• Field-proven performancen Nonelectric• Continuous operation, even with power
failuren S
afe• Suitable for Class I, Division 1 hazardous
locations

For complete specifications, refer to
Product Specification Sheet PSS 3-3A1 A.
Filled Thermal System Elements consist of sensor,
connection tubing, and mechanical element.

Functional Specifications
Thermal System Types:
Class IA:A thermal system that is completely filled with anincompressible liquid under pressure. The system
is fully compensated for ambient temperature
variations at case and along the tubing.

Class IIA:
A thermal system that is evacuated and partiallyfilled with a volatile liquid, such as methyl chloride(CL), ether (ET), butane (BU), or toluene (T).
Class IIIB:
A gas system that is filled under pressure withpurified nitrogen.

Standard Ranges, Standard Spans and Range Limits:				 CLASS IA							Range Limits		
Standard		
X Length
Spans				Minimum		Maximum°Cmm		 in°C		°C

CLASS IIA		
Standard
Ranges (Fills)
°C

25
122
4.8
13
120
38 to 105 (BU)
50
69
2.7
-15
200
105 to 150 (ET)
75
53
2.1
-38
200
50 to 205 (T)
100
43
1.7
-63
200
38 to 150 (ET)
150
36
1.4
-73
250
50 to 250 (T)
200
30
1.2
-73
260		
250
28
1.1
-73
260		
°F
mm		in°F		°F
°F
50
112
4.4
50
250
100 to 180 (CL)
100
69
2.7
0
400
100 to 220 (BU)
150
51
2.0
-50
400
100 to 250 (BU)
200
43
1.7
-100
400
100 to 270 (BU)
250
36
1.4
-100
400
100 to 300 (ET)
300
33
1.3
-100
500
100 to 350 (ET)
400
28
1.1
-100
500
100 to 400 (T)
500
25
1.0
-100
500
200 to 500 (T)							CLASS IIIB
Standard		 Range Limits
Spans
Minimum		Maximum°C°C		°C

100
150
200
250
300
400
500
°F
150
200
300
400
500
600
700
800
1000

-40
170
-15
400
-15
500
-15
500
-15
500
-15
500
-15
500
°F		°F
-50
240
-50
425
0
825
0
1000
0
1000
0
1000
0
1000
0
1000
0
1000

Page 59

Large Case Instruments (Elements)
THERMAL ELEMENTS

Performance Specifications
Accuracy:
Class IA:
±0.5% of calibrated span for spans up to 215°C(400°F).
±0.75% of calibrated span for spans between
215 and 330°C (400 and 600°F).
Class IIA:
±0.5% of calibrated span over upper 2/3 of scale.

Class IIIB:±0.5% of calibrated span for spans up to 330°C(600°F).
±0.75% of upper range value for upper range
value above 330°C (600°F).
±0.75% of lower range value for lower range
value below­45°C (-50°F).

Physical Specifications
Sensor Type:
Adjustable union, bendable extension.
Sensor Dimensions:
X = Sensitive portion length.
U = Insertion length.
J = Extension length.
B = Bushing thread type and size.
Y = Sensor diameter.

J = 450mm(18in).
X = 100 mm (4 in) for Class IIA and 150 mm (6 in)for Class IIIB systems. For Class IA systems, X
is governed by sensor diameter and measurement span. See table under “FUNCTIONALSPECIFICATIONS” on previous page.
U = (minimum) = X + 50 mm (2 in).
U = (maximum) = X + 380 mm (15 in).
B = 1/2, 3/4, or 1 NPT external threads as
specified for Class IA or IIA systems and 1
NPT external threads for Class IIIB systems.

Connecting Tubing:
1.5, 4.5, or 7.5 m (5, 15, or 25 ft) flexible
stainless steel over stainless capillary.

Y = 9.53 mm (3/8 in) for Class IA and IIA,
22.26 mm (7/8 in) for Class IIIB.

Optional Features
Alternative Thermal Systems:a) Longer capillary lenghts and types.
b) Other ranges and spans.
c) Sanitary fittings.
d) IIB system.

Page 60

Thermowells:
Refer to Page 35.
Mounting Flanges.
Special Purpose Sensors:
Preformed capillary, coated type, etc.

Large Case Instruments (Elements)
THERMAL ELEMENTS

How to Order1)	SPECIFY BASE INSTRUMENT BY MODEL NUMBER (Recorder, Indicator,
Controller, Transmitter)___________________________________________________________
2) SPECIFY CLASS/TUBING/SENSOR CODES
		 IA Liquid Filled: TA-1A/D-FS-SS/4542-J18______________________________________
		 IIA Vapor Filled: TA-2A/S-FS-SS/2542-J18______________________________________
		 IIIB Gas Filled: TA-3B/S-FS-SS/3542-J18________________________________________
3)	SPECIFY CALIBRATED RANGE (must meet requirements of table under“Functional Specifications” on Page 51)_____________________________________________
4) SPECIFY NORMAL WORKING TEMPERATURE______________________________________
5) SPECIFY AMBIENT TEMPERATURE AT CASE AND ALONG TUBING__________________
6) SPECIFY SENSOR DISTANCE ABOVE OR BELOW CASE (Class IIA systems only)_______
7) SPECIFY ANTICIPATED “U” DIMENSION (insertion length)___________________________
8)	SPECIFY BUSHING THREAD SIZE “B” DIMENSION; SPECIFY 1/2, 3/4, OR
1 NPT FOR IA OR IIA SYSTEMS, 1 NPT FOR IIIB SYSTEMS___________________________
9) SPECIFY OPTIONAL FEATURES___________________________________________________
10) SPECIFY INFORMATION FOR INSTRUMENT TAG___________________________________

Page 61Large Case Instruments (Elements)
SERVO ELEMENTS

Electronic Servo Elements
n Sealed Conductive Plastic Slidewire• Does not need frequent cleaning• Fast response and long lifen RFI Protected Electronics• Module may be field-mounted

n dc Torque Motor

Electronic Servo Elements are used with 40P
and 40M Series Recorders, Indicators, andControllers; 43AP and 43EP Series Indicating
Controllers; and 45P Series Transmitters.

• No brushes or gears

For complete specifications, refer to Technical
Information Sheet Tl 27-1a.

Functional Specifications
Standard Ranges:

Sensor Codes Available
Range		
Code
Range
mV mA
°C
°F
				
39
0 to 600
K			 T J		
Q										40—Measurement Range Code Table (see “HOW
TO ORDER” for description of sensor codes):Sensor Codes Available
Range		
Code
Range
mV mA
°C
01
0 to 5
M									020 to 10
M									030 to 20
M																040 to 30
Q			
05
0 to 50
M						Q										06
-50 to -50
Q							J						07-100 to +50
08
0 to 75
M						Q							J						
09-20 to +80
100 to 100°F

M			J			Q			

Q
Q

Q

					
11
-100 to 0
Q					
		
12-200 to +100
J E				 J E		
			E							13-100 to +100					
14
50 to 100
Q					
										
15-20 to +120
Q										
16—
17-25 to +125
T											J			Q					180 to 150Q		
19-50 to -150
J E																	2050 to 150
Q
21
4 to 20
A									2210 to 50
A											230 to 200
T J		 K Q		 J						E				24-100 to +200
J E						
25
100 to 200
Q				
		
26
0 to 250
T J			 Q T		 E
		T						J		27
-50 to +250						
28
50 to 250
Q		J		
			J		K29
0 to 300
Q T J									30
-350 to +300
T		E

Q
Q
Q
Q
Q

		J			Q		31
100 to 300
J			Q		J		K32
0 to 40
Q T J		K Q									
33100 to 40
K Q
		J			Q					34200 to 400
Q		J		K35
0 to 500
Q T J		K Q						
36
-150 to +500
T		E		
				37
200 to 500
K						
Q		J								38300 to 500

Page 62								
41200 to 600
J					J				
42300 to 600
T						J							430 to 700
Q			J					
44200 to 700
J		
			
45
0 to 750
J K					 K					R			J		460 to 800
Q								
47100 to 800
J K
				K				J
48300 to 800
K						
49
0 to 900
J				
Q
		K				J				
50
500 to 900
		K		S		J				
51
700 to 900
		K		S		J
52
0 to 1000
K									
53
500 to 1000
J K			
						
54
0 to 1200
J K			
						
55
200 to 1200
J K			
						
56
400 to 1200
J K			
			R							570 to 1300
										
58—				R						59800 to 1400
					60
900 to 1400
S					
				R			J			61
0 to 1500
							62500 to 1500
J K										K		630 to 1600
					
64
1100 to 1600
S		J											K		651200 to 1600
				R				
66500 to 1700
K									670 to 1800
J K		
								K		680 to 2000						
69
1000 to 2000
J		R		
								
701200 to 2000
R									K			71
0 to 2400							K			
72
1200 to 2400							K			73
0 to 2500							K			
74
1000 to 2500							K751500 to 2500
R S
									
761700 to 2500
S

Input Impedance (mA inputs):
4 to 20 mA (Range Code 21)-6.25 ohms.
10 to 50 mA (Range Code 22)-2.5 ohms.
Power Required:
See “HOW TO ORDER.”Large Case Instruments (Elements)
SERVO ELEMENTS

Performance Specifications
Accuracy:±0.5% of calibrated span.

Repeatability:
0.25% of calibrated span.

Optional Features
Loop Power Supply:
30 V dc.

Replacement Range Card:
Specify measurement type, measuring range code,
range units, and thermocouple burnout (if required).

dc Voltage Inputs:
0.055 to 42 V dc.

Power Failure Detection:
Positions pen either to 100% of range or to zero
when ac power fails.

Nonstandard Ranges:

Range
Range Limits
LimitsSpan Limits
Range
Limits

Item

lower

upper

minimum

maximum

Millivolts, Thermocouples
Resistance Temperature Detectors (RTDs)-15 mV
18 ohms

100 mV
330 ohms

5 mV
8 ohms

100 mV
300 ohms

Page 63

Large Case Instruments (Elements)
SERVO ELEMENTS

How to Order
1) SPECIFY BASE INSTRUMENT BY MODEL NUMBER:
2) SPECIFY SERVO ELEMENT: E______________­___ ___ ___ ___ ___Power Supply:

120Vac............................................................................................................................. S
220 V ac...........................................................................................................................N
240 V ac........................................................................................................................... P

Measurement—Sensor Type:

EC/ISA T (Cu-CuNi) Thermocouple......................................................................................T
DIN 43710-77 (Cu-CuNi) Thermocouple............................................................................ C
EC/ISA J (Fe-CuNi) Thermocouple.......................................................................................JDIN 43710-77 (Fe-CuNi) Thermocouple............................................................................ D
EC/ISA E (NiCr-CuNi) Thermocouple...................................................................................E
EC/ISA K (NiCr-NiAI) Thermocouple (DIN 43710-77)....................................................... K
EC/ISA R (Pt13Rh-Pt) Thermocouple....................................................................................R
EC/ISA S (Pt10Rh-Pt) Thermocouple (DIN 43710-77)........................................................S
EC/ISA B (Pt30Rh-Pt6Rh) Thermocouple............................................................................ B
dc mV ...................................................................................................................................... M
dc mA ...................................................................................................................................... A
Platinum RTD, 100-ohm DIN Resistance............................................................................. Q
Nonstandard (including Temperature Difference)............................................................ X

Measurement Range:

Select 2-digit number from Measurement Range Code Table on Page 54.......................... __­Range Units:

Degrees Celsius...................................................................................................................................... C
Degrees Fahrenheit.................................................................................................................................F
Millivolts.................................................................................................................................................... M
Milliamperes............................................................................................................................................ A

Burnout Feature (thermocouples only):Upscale action (not available with RTD)........................................................................................................ U
Downscale action (not available with RTD).................................................................................................. DNone ................................................................................................................................................................ N
Upscale action—Reverse Range (not available with RTD)........................................................................... R
Downscale action—Reverse Range (not available with RTD)...................................................................... S
None—Reverse Range...................................................................................................................................... T

3) SPECIFY OPTIONAL FEATURES_____________________________________________________
4) SPECIFY SENSOR (e.g., THERMOCOUPLE OR RTD) TO BE ORDEREDSEPARATELY______________________________________________________________________
5) SPECIFY INFORMATION FOR INSTRUMENT TAG____________________________________

Page 64

Large Case Instruments (Miscellaneous)Type 70 Electronic Contacts

70, MRn U
sed in 40P and 40M Series Recorders,

Indicators, and Controllers and 43AP
Series Indicating Controllers

Rotor Units:
Single or double rotor units are available, eachhaving 3 or 5 contact segments. One segment
serves as the common or collector ring
contact resulting in 2, 4, or 6 segments being
available for relay actuation. Contact segments
are adjustable with relation to each other by
micrometer adjustment screws. This relationshipis not disturbed by the contact setting
mechanism.
Contact Setting Mechanism:
Each rotor unit is independently positioned tooperate at desired values of measurement. A
wide selection of setting mechanisms is available.

A Type 70 Electric Contact System is used toactuate external control or alarm circuits.

MR Series Receiver Gauges
MR Series Receiver Gauges accept 3 to 15
psi inputs from remotely located pneumatictransmitters. A wide variety of dials and scales
is available to indicate measurements inconvenient engineering units.

Page 65

Panel-Mounted Pneumatic Instruments

Panel-Mounted Pneumatic Instruments
CONSOTROL 100 is a complete family of pneumatic controllers, recorders,
indicators, and auxiliary devices.
These instruments can handle the special requirements of virtually any application— from simple single control loops to interrelated multivariable systems.
Each CONSOTROL 100 unit is perfectly matched with other foxboro pneumatic
instruments — including transmitters, valves, and accessories.
And they are engineered to fit into control configurations of all sizes. Together,
they deliver exceptionally high system performance.
CONSOTROL 100 Instruments give you:

Bold, Readable Displays
Continuous scale indication of process variables and deviation from set point.

Separate Automatic and Manual Functions
Avoids forced shutdowns — control can be maintained in either mode.

Accurate Force Balance Control Modules
Metallic bellows, flexure-suspended balance member for precise opeation.

Balanceless, Bumpless Auto/Manual Transfer
Fast, safe selection of operating mode.

Simple, Low Cost Mounting
 helves accomodate up to ten instruments. Integral air manifold reduces pipingS
costs. Individual air shutoff valve for each instrument.

Easy Service
Pneumatic circuit board has simple piping and connections. Components arehighly accessible — all behind one sliding cover.

Page 66

Panel-Mounted Pneumatic Instruments (Indicating Controllers)
130M, 130P, 130F Series
Pneumatic CONSOTROL
Indicating Controllers

130M, 130P, 130Fn 130M—manual set point adjustment
n 130P—set point pneumatically adjusted

by a remote signal

n 130F—remote/local switch for manualor pneumatic adjustment of set point

For complete specifications, refer to
Product Specification Sheet PSS 2B-4H 1 A.

130M

130P

130F

130M, 130P, and 130F Series Pneumatic
CONSOTROL Indicating Controllers comparea remote transmitter signal with a set point andproduce a control signal to a final actuator.

Functional Specifications
Supply Pressure:
140 to 150 kPa or 20 to 22 psi.

Ambient Temperature Limits:-20 and +65°C (0 and 150°F) .

Input/Output Signal:
20 to 100 kPa or 3 to 15 psi, as specified.

Control Action Reversal:
The control action is reversed by a 90 degree
rotation of the reversing switch.

Mode Adjustment Limits:
Proportional Band: 5 and 500%.

Auto/Manual Transfer Unit:
A 2-position switch lever located on the frontof the manual unit is used to switch directlyfrom automatic to manual or, on controllers
with integral, from manual to automatic without
bumping the process.

Integral Action:
0.01 and 50 minutes per repeat.
Derivative Action:
0.01 and 50 minutes.

Performance Specifications
Accuracy:
Measurement and Set Point Indicators: ±0.5% ofcalibrated span.

Alignment Error:
Automatic Control Unit: Less than 0.5% ofcalibrated span.

Output Indicator:±2% of calibrated span.

Dead Band:
Automatic Control Unit: Less than 0.1 % of cali­brated span.

Repeatability:
Measurement and Set Point Indicators: Better
than 0.25% of calibrated span.

Measurement and Set Point Indicators: Less than
0.1% of calibrated span.

Physical Specifications
Mounting:
These controllers are designed for flush
mounting in vertical panels 3 to 25 mm (0.13
to 1 in) thick using 102 or 101 Series Shelves.
Each controller requires 1 unit of shelf capacity.
They may be calibrated for mounting in a panel
inclined up to 75 degrees from vertical with therear of the instrument lower than the front.

Connections:
Separate electric and pneumatic connectors arelocated on the rear of the chassis in accordance withthe standard arrangement for 102 Series Shelves.
Nameplate:White translucent plastic with 2.5 mm (0.1 in)high black letters. There can be a maximum of 2lines with 17 characters or spaces per line.

Page 67

Panel-Mounted Pneumatic Instruments (Indicating Controllers)
130M, 130P, 130F

Physical Specifications (continued)
Pointers:
Measurement: Red.
Manual Set (130M Series) and Remote Set (130P
and 130F Series):Black on white.
Local Set (130F Series): Black.
Output: Red.

Scales:Vertical Scale: Removable, transparent plastic withblack markings; 100 divisions, scaled as specified.
Output: Black markings on a white background;
20 divisions from 0 to 100% output signal. The redpointer continuously indicates automatic or manual
output. Refer to Chart and Dial Catalog 600.

Optional Features“Batch” Controller:
Prevents integral circuit saturation during
shutdown or other periods of sustained set
point and measurement deviation. Includes
internal adjustment for preload.

Reference BSL.

High “Batch” Unit:
Operates when controller output pressure
exceeds 100 kPa or 15 psi. Specify by
substituting 130MB, 130PB, or 130FB in place or
130M,130P, or 130F in Model Number.

External Set Point Connection:
Not available for 130F Series with External
Connection to Integral Bellows option. Provides
access to set point signal. Maximum transmission
distance is 1.5 m (5 ft) and all external connections
must be leaktight due to low air delivery of
aspirating relay. Use 1:1 repeating relay for
greater transmission distances.

Low “Batch” Unit:
Operates when controller output pressure falls
below 20 kPa or 3 psi. Specify by substituting
130MB, 130PB, or 130FB in place of 130M,
130P, or 130F in Model Number plus AS

External Connection to Integral Bellows:
Permits connection of external feedback signal toprevent integral circuit saturation.

Alarm Actuators and Alarm Lamps:
Refer to Page 72.

How to Order
1) SPECIFY MODEL NUMBER:________ __Controller:

with Manually Positioned Set Point............................................................................................130M-N
with Pneumatically Positioned Set Point.................................................................................... 130P-N
with Remote/Local Set.................................................................................................................. 130F-N

Control Function:

On/off ................................................................................................................................................................ 1
Proportional...................................................................................................................................................... 2
Proportional plus derivative........................................................................................................................... 3
Proportional plus integral............................................................................................................................... 4
Proportional plus integral plus derivative..................................................................................................... 52) SPECIFY INPUT/OUTPUT SIGNAL___________________________________________________
3) SPECIFY SET POINT AND MEASUREMENT SCALE RANGE—REFER TO CHART AND DIALCATALOG 600____________________________________________________________________
4) SPECIFY MOUNTING EQUIPMENT—102 OR 101 SERIES SHELVES_____________________
5) SPECIFY OPTIONAL FEATURES_____________________________________________________
6) SPECIFY INFORMATION FOR NAMEPLATE AND INSTRUMENT TAG___________________

Page 68

Panel-Mounted Pneumatic Instruments (Indicating Controllers)
130 Series Pneumatic
CONSOTROL Auto
Selector Control System

130 AUTO-SELECTORn One Final Actuator• To prevent any of several process measurements from exceeding their set points

n Continuous Measurement• Variables are measured and control is 		transferred to the one that is tending to 		
exceed its limitn Auto/Manual Switch• Smooth transfer by means of a single 		2-position switch• Balanceless, bumplessn A Complete System Consists of:• One 130MS (local set) or 130FS (local/
remote set) controller with manual transfer unit
• One or more 130MA (local set) or 130FA (local/remote set) controllers without manual units• Rack-mounted high or low selector relays• A rack-mounted seal valve

The 130 Series Pneumatic CONSOTROL AutoSelector Control System controls two or more
related variables within preset limits.

For complete specifications, refer toGeneral Specification Sheet GS 2B-4H 1 D.

Functional Specifications
Supply Pressure:
140 to 150 kPa or 20 to 22 psi.

Ambient Temperature Limits:-20 and +65°C (0 and 150°F) .

Input/Output Signal:
20 to 100 kPa or 3 to 15 psi, as specified.

Control Action Reversal:
The control action is reversed by a 90 degree
rotation of the reversing switch.

Mode Adjustment Limits:
Proportional Band: 5 and 500%.

Auto/Manual Transfer Unit:
A 2-position switch lever located on the frontof the manual unit is used to switch directlyfrom automatic to manual and from manual toautomatic without bumping the process.

Integral Action:
0.01 and 50 minutes per repeat.
Derivative Action:
0.01 and 50 minutes.

Performance Specifications
Accuracy:
Measurement and Set Point Indicators: ±0.5% ofcalibrated span.

Dead Band:
Automatic Control Unit: Less than 0.1% ofcalibrated span.

Output Indicator: ±2% of calibrated span.

Measurement and Set Point Indicators: Less than
0.1% of calibrated span.

Repeatability:
Measurement and Set Point Indicators: Better
than 0.25% of calibrated span.

Alignment Error:
Automatic Control Unit: Less than 0.5% ofcalibrated span.

Physical Specifications
Mounting:
These controllers are designed for flush
mounting in vertical panels 3 to 25 mm (0.13
to 1 in) thick using 102 or 101 Series Shelves.

Each controller requires 1 unit of shelf capacity.
They may be calibrated for mounting in a panel,
inclined up to 75 degrees from vertical, with therear of the instrument lower than the front.

Page 69

Panel-Mounted Pneumatic Instruments (Indicating Controllers)
130 AUTO-SELECTOR

Physical Specifications (continued)
Connections:
Separate electric and pneumatic connectors arelocated on the rear of the chassis in accordancewith the standard arrangement for 101, 102
Series Shelves..

Output:Black markings on a white background; 20 divisions
from 0 to 100% output signal. The red pointercontinuously indicates automatic or manual output.
Refer to Chart and Dial Catalog 600.

Nameplate:White translucent plastic with 2.5 mm (0.1 in)high black letters. There can be a maximum of 2lines with 17 characters or spaces per line.

Nominal Dimensions (including manual unit):
205 mm high x 70 mm wide x 555 mm deep (8.2
in high x 2.8 in wide x 21.8 in deep).

Output: Red.

Selector Relays and Volume Booster:
A low selector relay is required for eachcontroller in a “LOW” Auto-Selector System. A“HIGH” selector system requires one less highselector relay than the number of controllers
and one low selector relay used as a volume
booster in the system output. Specify High
Selector, Part No. B0114YL or Low Selector, Part
No. B0114BZ.

Scales:Vertical Scale: Removable, transparent plastic
with black markings; 100 divisions, scaled as
specified.

Seal Valve:
One pneumatically actuated seal valve isrequired per Auto-Selector System. Specify Part
No. C0138RM.

Pointers:
Measurement: Red.
Manual Set (130M Series) and Remote Set (130F
Series): Black on white.
Local Set (130F Series): Black.

Optional Features“Batch” Controller:
High “Batch” feature available only forcontrollers used in “LOW” Auto-Selector
Systems. Add Suffix -B to Model Number.

Alarm Actuators and Alarm Lamps:
Refer to Page 72.

How to Order
1) SPECIFY MODEL NUMBER:________ _____Controller:

with Manually Positioned Set Point.................................................................................... 130M
with Remote/Local Set Point.................................................................................................130F

Auto-Selector:

without Manual Transfer Unit.............................................................................................................. Awith Manual Transfer Unit.................................................................................................................... S

Control Function:

Proportional plus integral.............................................................................................................................N4
Proportional plus integral plus derivative...................................................................................................N5

2) SPECIFY INPUT/OUTPUT SIGNAL___________________________________________________
3) SPECIFY SET POINT AND MEASUREMENT SCALE—REFER TO CHART AND DIALCATALOG 600____________________________________________________________________
4) SPECIFY SELECTOR RELAYS AND SEAL VALVE______________________________________
5) SPECIFY MOUNTING EQUIPMENT—102 OR 101 SERIES SHELVES_____________________
6) SPECIFY OPTIONAL FEATURES_____________________________________________________
7) SPECIFY INFORMATION FOR NAMEPLATE AND INSTRUMENT TAG___________________

Page 70

Panel-Mounted Pneumatic Instruments (Indicating Controllers)
110 Series Pneumatic
CONSOTROL Indicators

110

n Simple Construction• No cables, clutches, or complicated 		mechanisms• Ball and socket connectors for 			frictionless linkagen Bold, Readable Display• Large, brightly colored pointers

n Easy Calibration• Indicator partially slides out of shelf• Adjustments made at front end of paneln Integral Recording• Optional circular chart recording 		mechanism

n Versatile Mounting• Singly or with other instruments in 102 		or 101 Series Shelves

The 110 Series Pneumatic CONSOTROL
Indicators indicate 1, 2, or 3 process
measurements received as proportional
pneumatic signals from remote transmitters.

For complete specifications, refer toGeneral Specification Sheet GS 2B-3D1 A.

Functional Specifications
Element:
One brass bellows per variable.

Scale:
Vertical. Effective length 100 mm (4 in). Blackfigures on clear plastic, scaled as specified.

Damping:
Integral and adjustable.
Input Signal:
20 to 100 kPa or 3 to 15 psi, as specified.

Pointers:
Red, green, blue.
Ambient Temperature Limits:-20 and +65°C (0 and 150°F).

Performance Specifications
Accuracy:±0.5% of calibrated span.

Dead Band:
Less than 0.1 % of calibrated span.

Repeatability:
0.25% of calibrated span.

Physical Specifications
Mounting:
Flush in panels 3 to 25 mm (0.13 to 1 in) thickusing 102 or 101 Series Shelves. Indicators
require 1 unit width of mounting capacity.

Nameplate:White translucent plastic with 2.5 mm (0.1 in)high black characters. Maximum of 2 lines, 17characters or spaces per line.

Connections:Electrical and pneumatic multiprong receptacles
on rear of chassis.

Overall Dimensions:
165 mm high x 71 mm wide x 554 mm deep (6.5
in high x 2.8 in wide x 21.8 in deep).

Page 71

Panel-Mounted Pneumatic Instruments (Indicating Controllers)
110

Optional Features
Chart Record:
Single variable indicators only may be furnishedwith an integral recording mechanism toprovide a continuous record on a nominal 125
mm (5 in) circular chart with 50 mm (2 in) wide
recording area. Electrical 2-speed chart drive,
24 h/24 min/off or 7 day/2.8 h/off rotation, as
specified. Supply voltage 24, 120, or 240 Vac, 50 or 60 Hz, as specified. 24 V ac must be
specified when alarm option is used. Foxboro
and CSA certified for ordinary and Class I,
Groups C and D, Division 2 hazardous locations.
CSA certified units, Source - Canada. Max­imum
angle mounting is 15 degrees from vertical.

Alarm Actuators and Alarm Lamps:
Refer to Page 72.

How to Order
1) SPECIFY MODEL NUMBER:________One Variable.............................................................................................................................................. 111
Two Variables............................................................................................................................................ 112
Three Variables.......................................................................................................................................... 113

2) SPECIFY INPUT SIGNAL____________________________________________________________
3) SPECIFY SCALE RANGES—REFER TO CHART AND DIAL CATALOG 600________________
4) SPECIFY MOUNTING EQUIPMENT—102 OR 101 SERIES SHELVES_____________________
5) SPECIFY OPTIONAL FEATURES_____________________________________________________
6) SPECIFY NAMEPLATE AND INFORMATION FOR INSTRUMENT TAG___________________

Page 72

Panel-Mounted Pneumatic Instruments (Recorders & Controllers)
127S Series Pneumatic
CONSOTROL Recorders

127S

n Bold, Readable Displays• Bright red, green, and blue ribbon 		
indicators• Individual scalesn Simple Maintenance• Accessible modular components• Adjustments can be made while 		instrument is in operationn Easy Chart Changing and Pen

Replacement• Removable chart drive• Snap-in, fiber-tip pen cartridges

n Precise Servo Mechanism• Accurately positions pens
• All-pneumatic• Guided by force balance detector

The 127S Series Pneumatic CONSOTROL
Recorders record up to 3 process
measurements received from remotetransmitters.

For complete specifications, refer to
Product Specification Sheet PSS 2B-3E2 B.

Functional Specifications
Pneumatic Input Signal:
3 to 15 psi (20 to 100 kPa) (0.20 to 1.0 Kg/cm2)
or 3 to 18 psi (20 to 120 kPa) (0.2 to 1.2 Kg/cm).
Supply Pressure:
120 to 150 kPa or 18 to 22 psi.
Power Requirements:
24 ±4 V ac, 120 or 240 V ac, +10, -15%, 50 or
60 Hz, as specified. Required with electric chart
drive only.
Fusing:Required with electric chart drive only. For 24 V
ac: No fuse required.
For 120 V ac: 3/8 A.
For 240 V ac: 1/4 A.
Ambient Temperature Limits:
5 and 50°C (40 and 120°F) .

Pen Speed:
The elapsed time for the pen to travel from 10
to 90% of full scale is adjustable from less than 3
to greater than 20 seconds.
Mounting Position:
Normal mounting position is horizontal.
Instrument can also be mounted up to a 15
degree angle with the rear of the instrument
above the front, or down to a 75 degree angle
with the rear of the instrument below the front.
Adjustments:
Zero and span adjustments are accessible inthe service (partially withdrawn) position, andprovide a minimum of ±2% adjustment after
the instrument has been calibrated. Ribbon
indicator has an adjustment for zeroing relative
to pen. No separate ribbon span adjustment isnecessary, since the ribbon travels with the pen.
Electrical Classification:
These instruments have been designed for use
in ordinary locations.

Performance Specifications
Accuracy:±0.5% of calibrated span.

Repeatability:
0.15% of calibrated span.

Page 73

Panel-Mounted Pneumatic Instruments (Recorders & Controllers)
127S

Physical Specifications
Connections:
Separate electric and pneumatic connectors arelocated on the rear of the instrument chassis inaccordance with the standard arrangement for
102 and 101 Series Shelves.
Front Door:
Hinged at bottom to provide access to chartand pens. Lens material is transparent molded
polycarbonate.
Right Side Access Cover and Calibration Door:
Removable access cover and a hinged-down
calibration door provide access to internal parts
and adjustments.
Nominal Dimensions:
69 mm wide x 165 mm high x 478 mm long (2.7
in wide x 6.5 in high x 18.8 in long).
Pens:
1, 2, or 3, as specified.
Red Pen: Located in center position.

Left Side Access Cover:Sliding cover provides access to internal
parts. Cover material is transparent molded
polycarbonate.
Instrument Mounting:Up to 10 instruments can be mounted in
Foxboro 102 Series Shelves (706 mm [27.8 in]deep). Up to 4 instruments can be mounted in
Foxboro 101 Series Shelves (544 mm [21.4 in]deep). Shelves are designed for flush mounting
in 3 to 25 mm (0.13 to 1 in) thick panels.
Nameplate:
A gray polycarbonate nameplate is attachedto the lower surface of the front door. It
provides space for 3 lines which can haveapproximately 24 characters (including spaces)
to accommodate a description of each input
signal. A green, red, and blue (from top tobottom) color identification dot at the beginningof each line corresponds to the pen color andapplicable input signal description.

Green Pen: Located in inner position.
Blue Pen: Located in outer position.
Pen Separation: Approximately 12 minutes atthe standard chart speed of 20 mm/h (0.8 in/h).

Optional Features
Chart Drives:
The following chart drives are available at supply
voltages of 24, 120, or 240 V, and frequencies
of 50 or 60 Hz; EXCEPTION: 5 mm/h (0.2 in/h)
not available with 24 V. Selectable by specifying
single or dual speed, chart speed, voltage, andfrequency.
Single Speed: 5, 10, 20, 40, 80, 120, 320, 720
and 2400 mm/h (0.2, 0.4, 0.8, 1.6, 3.1, 4.7, 6.4,
28.2 and 96.0 in/h).
Dual Speed: 20 mm/h and 20 mm/min (0.8 in/h
and 0.8 in/min). 20 mm/h and 40 mm/min (0.8in/h and 1.6 in/min).
Pneumatic Impulse: 20 mm/h (0.75 in/h).

Page 74

Special Pen Position for 2-Pen Recorder:
Optionally available is a 2-pen recorder with thered pen in the center position and the blue penin the outer position. Selectable by specifying
model number for 2-pen recorder and AS
Reference SPP.
Charts:A selection of standard charts is available. Refer
to Foxboro Chart and Dial Catalog 600.
Alarm Lamps:
Refer to Page 72.

Panel-Mounted Pneumatic Instruments (Recorders & Controllers)
127S

How to Order
1) SPECIFY MODEL NUMBER: 127S-______ __ – __Pen and Indicator Quantity and Color:

1 pen and 1 indicator (red)........................................................................................... 1
2 pens and 2 indicators (red and green).................................................................... 2
3 pens and 3 indicators (red, green, and blue)......................................................... 3

Chart Drive:

Electric, single speed, 20 mm/h (0.8 in/h)...........................................................................E
Pneumatic impulse, 20 mm/h (0.8 in/h). Not available with code 3 above.
		 Codes X and 0 below must be specified.......................................................................P

Voltage:

120 V ac............................................................................................................................................A
240 V ac............................................................................................................................................ B
24 V ac..............................................................................................................................................C
None. Select with Chart Drive code P..........................................................................................X

Frequency:

50 Hz ....................................................................................................................................................... 5
60 Hz ....................................................................................................................................................... 6
None. Select with Chart Drive code P.................................................................................................. 0

Optional Feature:

3 alarm lamps, externally operated. Available with Voltage code C above only................................... L

2) SPECIFY INPUT SIGNAL____________________________________________________________
3) SPECIFY SCALE RANGES AND CHART RANGES FROM CHART AND DIALCATALOG 600____________________________________________________________________
4) SPECIFY MOUNTING EQUIPMENT—102 OR 101 SERIES SHELVES_____________________
5) SPECIFY OPTIONAL FEATURES_____________________________________________________
6) SPECIFY INFORMATION FOR NAMEPLATE AND INSTRUMENT TAG___________________

Page 75

Panel-Mounted Pneumatic Instruments (Loading, Switching & Bias)
135 Series Manual
Loading, Switching, andBias Stations

135

n Selective Pneumatic Switchingn Remote Control Settingn Manual Loading of Final Operators

135 Series Stations are complementary
auxiliary units for 100 Series Control Systems.

For complete specifications, refer toGeneral Specification Sheet GS 2B-4H2 A.

Functional Specifications
Model 135S:
Passes a standard input signal to a final operator
or manually generates an accurate standard signal.
It is normally used in control loops where a single
master controller regulates two or more final operators. The station contains a vertical scale, indicating the input signal and the output of the manual
generator. By matching the two indicators on thevertical scale and flipping the simple 2-positiontransfer switch, a bumpless transfer is ensured.
Model 135SG:
Performs the same function as the Model 135S
(above). An additional pointer is provided todisplay a standard input signal.

Model 135T:
Provides an accurate manually adjusted standard
output signal used for remote manual control of
final operators or for the remote setting of any
pneumatic receiver. A manual adjusting knob
regulates a stable standard output indicated on
a 100 mm (4 in) vertical scale.
Model 135TG:
Performs the same function as the Model 135T
(above). An additional pointer is provided todisplay a standard input signal.
Model 135TG2:
Performs the same function as the Model 135T.
Two additional pointers are provided to display
standard input signals.

Page 76

Model 135B:
Performs the same functions as the Model 135S
with the added feature that up to ±100% bias may
be manually added to the standard input signal. It
is normally used where a single master controller
regulates two or more final operators, or where it
remotely sets two or more controllers, and where itis necessary to bias one with respect to the other.
This unit has three pointers. One is used to indicate
the manually generated signal and the other two
indicate the input signal before and after it is biased.
Ambient Temperature Limits:-20 and +65°C (0 and 150°F).
Input/Output Signal and Supply Pressure:
Input/Output Signal

Supply Pressure

20 to 100 kPa
3 to 15 psi

140 to 150 kPa
20 to 22 psi

Panel-Mounted Pneumatic Instruments (Loading, Switching & Bias)
135

Performance Specifications
Receiver-lndicator:
Accuracy: ±0.5% of calibrated span.

Manual Output Pointer:
Output Indicator: 0.5% of calibrated span.

Repeatability: 0.25% of calibrated span.

Repeatability of Setting: 0.25% of calibrated span.

Dead Band: Less than 0.1% of calibrated span.

Physical Specifications
Element:
Brass bellows.
Scale:
100 mm (4 in) vertical, black markings on whitebackground.
Pointers:
Manual output: black.
Bias output: red.
Inputs: green and blue.

Connections:Electrical and pneumatic multiprong receptacles
on rear of chassis.
Nameplate:White translucent plastic with 2.5 mm (0.1 in)high black letters. Maximum of 2 lines, 17 lettersor spaces per line.
Mounting:
Flush in panels 3 to 25 mm (0.13 to 1 in) thickusing 102 or 101 Series Shelves. Units require 1
unit of mounting capacity.

Optional Features
Alarm Actuators and Alarm Lamps:
Refer to Page 72.

How to Order
1) SPECIFY MODEL NUMBER:________

Manual Loading........................................................................................................................................ 135T
Manual Loading with Variable Indicator..............................................................................................135TG
Manual Loading with Two Variable Indicators................................................................................... 135TG2
Automatic/Manual Switching.................................................................................................................. 135S
Automatic/Manual Switching with Variable Indicator........................................................................135SG
Automatic/Manual Switching and Bias.................................................................................................. 135B

2) SPECIFY INPUT AND OUTPUT SIGNALS_____________________________________________
3) SPECIFY SCALE RANGES—REFER TO CHART AND DIAL CATALOG 600________________
4) SPECIFY MOUNTING EQUIPMENT—102 OR 101 SERIES SHELVES_____________________
5) SPECIFY OPTIONAL FEATURES_____________________________________________________
6) SPECIFY INFORMATION FOR NAMEPLATE AND INSTRUMENT TAG___________________

Page 77

Panel-Mounted Pneumatic Instruments (Subpanel)
PM960 Pneumatic Subpanel

PM960

n P
neumatic circuitry cast into base platen Many versions available, including

Pneumatic Subpanels are used in conjunction
with pneumatic Standard Controllers forgenerating set point, manual output, and forauto-manual transfer switching.

VersionFront View

Manual Loader with process variableindicatorn Set point control knob secures to prevent
accidental alteration
Circuit Diagram

Pneum. Connections

Manual Loader-A

1 Zul. Supply air
5yH Manual output

For Local Set
Control-F

1 Zul. Supply air
2S
Switch signal
3w Setpoint
4yA Controller output
5Y
OutputPneumatic Subpanel
Front finish........................ bezel grey RAL 7037
facia grey RAL 7035
Supply air.......................... 1.4± 0.1 bar / 20 ± 1.4 psi
Manual-Auto-switch........ flat slide valve

Pneumatic Signal Generator (built in)
Output adjustable.............0 to supply air pressure
Air consumption................≤ 100 l/h per signal gen.
Supply air dependency....≤ 0.2 %/0.1 bar
Temperature effect...........≤ 0.3 %/10 K

Pressure Indicator (built in)
Version.............................. single-or double indicator
System............................... bourdon tube
Indication range.............. 0 to 1.2 bar / 0 to 18 psi
Scale length..................... 44mm
Error limit.......................... ≤ 1 % f. s. d.
Cross-matching error..... ≤ 0.5 % f. s. d.
(Double indicator)

Materials
Bourdon tube.....................bronze
Other components...........al, brass, zinc-plated steel

Page 78

For complete specification, refer to ProductSpecification Sheet PSS EIC4315 A-(en)Panel-Mounted Pneumatic Instruments (Computing Elements)
PM960 , 556, 557

How to Order			
Pneumatic Subpanel
Version:___ ___ ___PM960

Manual Loader................................................................................................................................ -A
Manual Loader with process indicator........................................................................................ -B
Double Manual Loader................................................................................................................. -C
for slave control.............................................................................................................................. -D
for Slave control, controller with PI switch.................................................................................. -E
for local set control........................................................................................................................ -F

Indicating Range (Scale):0-1.2bar.................................................................................................................................................... 1
0-18psi...................................................................................................................................................... 2
0-120kPa................................................................................................................................................... 3

Options:

Wall mounting plate......................................................................................................................................-W
Tag No. Labeling with inscription.................................................................................................................-L

556 Series and Model 557
Pneumatic Analog Computersn Versatile• Multiply• Divide• Square• Square rootn W
 ide Range of Applications• Flexible scaling for multiplication and 		
divisionn Built-in Function Switch• To select any of four computing 		functionsn Weatherproof• Plastic cover• Die-cast, low copper content aluminumalloy manifold• IEC IP53 and NEMA Type 3
For complete specifications, refer to ProductSpecification Sheets PSS 2B-5E1 A and B.

Units provide output proportional tomathematical function of input. Model 557
performs square root function only.

Functional Specifications
Functions and Related Equations:
The 556 Series performs any one of fourfunctions and solves an equation as shown in
the table. The 557 performs square root only.
Function

Equation

Multiplication

A = fB(z+sC)AB=(f) (z + sC)

Division
Square RootB = √A

SquareA = C2A, B. and C are pneumatic signals expressed asdecimal equivalents of percent of span.
f = Scaling factor (limits: 0.750 and 3.500).
s = Span factor for C signal (limits: 0.050 and
1.000).
z = Zero suppression of C signal (limits: 0.000
and 0.950).
NOTE: The sum of s and z must equal 1.000.

Input and Output Signals:
20 to 100 kPa or 3 to 15 psi, as specified.
Supply Pressure Limits:
130 and 150 kPa or 19 and 22 psi.
Ambient Temperature Limits:-40 and +80°C (-40 and + 180°F).
Air Consumption Under Normal Operation:<.10 m3/h (.06 scfm) at standard conditions.
Air Connections:
All air connections are tapped for 1/4 NPT.
Mounting:
A bracket is supplied for vertical mounting on arack or DN 50 or 2 in pipe. The bracket is designed
for either horizontal or vertical pipes or rack struts.
Page 79

Panel-Mounted Pneumatic Instruments (Computing Elements)
556, 557

Performance Specifications
Accuracy:
Multiplication: ±0.5% of output span referred to
input.

Repeatability:
0.1 % of output span.

Division: ±0.5% of output span referred to input.
Square Root:
556: ±0.25% of output span referred to input.
557: ±0.5% of output span referred to input.
Square: ±0.5% of output span.

Physical Specifications
Materials of Construction:
Cover: Molded plastic with textured gray finish.
Manifold: Low copper content aluminum alloy
die-casting with a gray epoxy finish.

Environmental Protection:
The housing construction is weatherproof. It
is dust-protected as defined in IEC IP53 andprovides the environmental protection of NEMA
Type 3.

Flexures: 17-4 PH stainless steel.
A and C Bellows: Brass.
B Diaphragm: Copper alloy.

How to Order
1) SPECIFY MODEL NUMBER:
Function

FormulaMultiplication

A = fB(z+sC)

Division

B = A/f(z + sC)

Square Root (a)B = √A

Squarea + C2

Square RootB = √A________­.............................................................................556-8.............................................................................556-9........................................................................... 556-10........................................................................... 556-11.............................................................................. 557(a) Refer also to Model 557, and accuracy statement in “PERFORMANCE SPECIFICATIONS.”

2) SPECIFY INPUT/OUTPUT RANGE IN PROCESS UNITS_________________________________
3) SPECIFY FOR MODELS 556-8 AND 556-9:
SUPPLY THE VALUE OF f, s, AND z__________________________________________________
DEFINE INPUT/OUTPUT SIGNAL RANGE DESIRED, kPa OR psi________________________
4) SPECIFY INFORMATION FOR INSTRUMENT TAG____________________________________

Page 80

Panel-Mounted Pneumatic Instruments (Computing Relay)
PC191 PneumaticComputing Relay

PC191

n Pneumatic Subpaneln Time Function Relays• Booster Relay• Selector Relay• Reversing Relay• Adding Relay• Averaging Relay• Subtraction Relay• Fixed Ratio Relay• Square root
extractor• Function generator• Delay Relay• PI Relay• PD Relay• Delay element• Trend Relay

For complete specification, refer to ProductSpecification Sheet PSS EIO4332 A-(en)How to Order					____ ___
Pneumatic Computing Relay
PC191
Function and Formula:

Booster Relay A = E............................................................................................................................ -100
Min.-selector relay 2 inputs................................................................................................................ -101
Max.-selector relay 2 inputs............................................................................................................... -102
Reversing relay A = K - E.................................................................................................................... -110
Booster Relay A = E with calibration adjustment............................................................................ -113
Adding relay A = E1+E2-K................................................................................................................. -121
Averaging relay A = (E1+E2) / 2....................................................................................................... -130
Subtraction relay A = E1-E2 +K......................................................................................................... -140
Fixed ratio relay A = 0.5E - K............................................................................................................. -221
Fixed ratio relay A = 0.5E + K............................................................................................................ -231
Fixed ratio relay A = 2E - K................................................................................................................ -261
Fixed ratio relay A = 3E - K................................................................................................................ -281
Square root extractor A = E(xp=1/2) 0.2-1.0 bar............................................................................ -611
Square root extractor A = E(xp=1/2) 3-15 psi................................................................................. -612
Function generator A = F (E) 0.2-1.0 bar......................................................................................... -621
Function generator A = F (E) 3-15 psi.............................................................................................. -622
Delay relay symmetrical..................................................................................................................... -900
Delay relay signal increase................................................................................................................ -901
Delay relay signal decrease............................................................................................................... -902
PI-relay.................................................................................................................................................. -905
PD-relay................................................................................................................................................ -910
Delay element (without booster)...................................................................................................... -913
Trend relay V = 2.5............................................................................................................................. -920

Options:

Working on cams with given function (only for -621 / -622).................................................................... -A

Page 81Panel-Mounted Pneumatic Instruments (Computing Relay)
PS953

PS953 Pneumatic Signal
Selection Relayn Signal selector relays• Signal selector Relay with mechanicalor pneumatic setpoint adjustment forpressure and pressure differencen Relays for control applications• Lock-in relay• Limiting relay• Changeover relay• Push-button valve

For complete specification, refer to
Product Specification Sheets PSS
EIO4331 A-(en) for PS953

How to Order					____ ___
Pneumatic Signal Selection Relay
PS953
Function:

Signal selector relay, mechanical setpoint change-over if setpoint is exceeded...................... -100
Signal selector relay, mechanical setpoint change-over if setpoint is drops............................. -101
Three-way signal selector relay, mech. setpoint, change-over if setpoint is exceeded,		 reversing if setpoint is drops........................................................................................................ -111
Three-way signal selector relay, mech.setpoint (0,2-1,0 bar) with self-latch,		 change-over if setpoint is exceeded reversing manual (a)...................................................... -112
Three-way signal selector relay, mech.setpoint (0-1,0 bar) with self-latch,		 change-over if setpoint is exceeded, reversing manual (a)..................................................... -114
Lock-in relay single-action................................................................................................................. -200
Lock-in relay double-action............................................................................................................... -210
Limiting relay for max.pressure......................................................................................................... -410
Change-over relay pneum. conn. 1/8 - 27 NPT.............................................................................. -611
Change-over relay pneum. conn. 1/4 - 18 NPT.............................................................................. -613

Options:

Push-button valve for venting pneum. signal lines (only for -112/114).................................................. -A
Adapter Plate for PS953-210 for large pneumatic connector..................................................................M
Adapter Plate for PS953-200 for NAMUR mounting instead of SOV......................................................NNOTES:(a) With Option - A(b) With Option - B

Page 82

Panel-Mounted Pneumatic Instruments (Miscellaneous)
130MD, 130PD, 130FD, 138, 130M-N, 102, 101, 120

130MD, 130PD &130FD Series Pneumatic
CONSOTROL Control Stations
130MD, 130PD, and 130FD Series
Pneumatic CONSOTROL Control
Stations provide Panel-Mounted
measurement, set point, andauto/ manual transfer functions
to operate separately mounted
controllers. The stations are usedwith the 138 Series Controllers tominimize time lag resulting fromlong distances between control
panels and the process.

130M-N Series Pneumatic
CONSOTROL Indicating
Gap Controllers

138 Series Remote Mounted
Pneumatic Controllers

138 Series Remote Mounted PneumaticControllers are used with 130MD,130PD, and130FD Series Control Stations. They provide theresponse advantage of a local controller withthe convenience of a Panel-Mounted display.

102 & 101 Series Pneumatic
CONSOTROL Shelves

The Model 130M-N7 Differential
Gap Controller is an on/off unit
with an adjustable inert zone (gap)within which the controller will not
respond to changes in the measurement.
The Model 130M-N94 Gap Action
Floating Controller provides standard proportional plus integral control action when the measurement
exceeds the high or low gap limit.

120 Series Pneumatic
CONSOTROL Recorders

120 Series Pneumatic CONSOTROL Recorders
record up to four process measurements received
as standard pneumatic transmission signals fromremote transmitters. Large, brightly colored pensclearly indicate the measurement. The recording
is on a SCAN-FOLD chart.

102 and 101 Series Pneumatic CONSOTROL
Shelves allow flush panel mounting of pneumatic
CONSOTROL instruments in panels. They arecompact units, available in ten sizes for one to ten
instruments. The shelves allow instruments to bewholly or partially withdrawn without interferingwith the operation of others.

Alarm Actuators
and Alarm Lamps

100 Series Pneumatic Alarm Actuators areavailable integrally mounted in CONSOTROL 100
instruments. The actuators are operated directlyby the pneumatic input signals. They provide
high sensitivity, accuracy, and dependability byeliminating intermediate receiver mechanisms. A
wide variety of alarm configurations is available.
Page 83

Converters

Converters
Many processes require both electronic and pneumatic instrumentation.
Converters allow you to integrate that combination into your control strategy.

Foxboro makes converters for both purposes:
CURRENT-TO-PNEUMATIC (I/P) E69 SERIES• Available with a selection of input/output signals• Small, compact, lightweight—for easy installation• High air delivery rate for direct valve actuation

CURRENT-TO-PNEUMATIC (I/P) IP24 SERIES• Available with a large selection of input/output signals• Designed for field service / heavy duty application• Stainless steel version as option

Page 84

Converters (Current-to-Pneumatic)
E69, E69R

E69F and E69R Series
Current-to-Pneumatic
Converters
n Light, Compact• Easy to installn Low Air Consumption• Reduced operation costsn High Air Capacity• Eliminates need for volume boosters
n Choice of Various Input and Output Signals• Suitable for all applicationsn Sealed Air Purged Case (E69F)• Long-term dependability in dirty 		environments
n Precise Performance

E69F and E69R Series Current-toPneumatic Converters accept a dc
milliampere input signal and convert to a
proportional pneumatic output signal.

For complete specifications, refer to
Product Specification Sheets PSS 4-8B1 A(E69F) and 2B-2A2 A (E69R).

Functional Specifications
Input Signal Ranges (mA dc):
4 to 20, 4 to 12, 12 to 20.
10 to 50, 10 to 30, 30 to 50.

Supply Pressure: (a)Output Signal Ranges:
20 to 100 kPa or 3 to 15 psi.
40 to 200 kPa or 6 to 30 psi.
3 to 27 psi.
7 to 125 kPa or 1 to 18 psi.
7 to 220 kPa or 1 to 32 psi.

NominalLimits140 kPa or 20 psi
240 kPa or 35 psi

130 and 160 kPa or 19 and 23 psi
225 and 260 kPa or 33 and 38 psia) Not less than 20 kPa or 3 psi above maximum output.

Electrical Classification:
FM, CSA, and ATEX certified. Explosion proof
and intrinsically safe versions available. Refer toFoxboro for complete specifications.

Input Impedance:
170 Ω (4 to 20 mA dc).
27 Ω (10 to 50 mA dc).

Physical Specifications
Accuracy:
Output Signal codes 2 and 3: ±1.0% of
calibrated span.
Output Signal codes 7 and 8: ±2% ofcalibrated span.
Linearity:
0.5% of calibrated span typical.
Repeatability:
0.25% of calibrated span typical.

E69F Enclosure:

Die-cast copper (1% maximum) aluminumalloy body and cover (ASTM 512A and 512B),
with a vinyl or epoxy powder finish. Meets therequirements of IEC IP65 and provides theenvironmental protection of NEMA Type 4.
Mounting:
Valve yoke, surface, or DN 50 or 2 in pipe.
Calibrated for vertical mounting unless otherwise
specified. Bracket is supplied as standard.
Input/Output Connections:Electrical: Tapped 1/2 in for conduit fitting toconnect to a pair of 0.5m (18 in) long 18 AWG(1.0 mm2) twisted leads.
Pneumatic:
1/4 NPT for air supply and output signal.

Page 85

Converters (Current-to-Pneumatic)
E69F, E69R

Physical Specifications (continued)
E69R
Enclosure:
Die-cast aluminum front and back plates withextruded top and bottom rails and steel side
covers. Finish is vinyl, polyurethane, or
acrylic enamel.

Mounting:
Up to 6 units may be mounted across in a 483
mm (19 in) rack.
Input/Output Connections (on front panel):
Electrical: Screw terminal with terminal cover.
Pneumatic: 1/4 NPT for air supply and out signal.

Optional Features
Alternative Output Signal:
Refer to Foxboro for other than standard output
signals.

Mounted integral to converter. PG 11 Cable
Gland (Trumpet Connector) for non explosionproof applications.

Adjustable Filter Regulator (E69F):
Provided without gauge.

Miniature Junction Box (E69F):
Consists of an aluminum box, 64 x 58 x 33 mm(2.5 x 2.3 x 1.3 in), and meets the IEC/ IP65
requirements and the environmental protectionof NEMA Type 4. Provided with 1/2 in conduit,
20 mm threaded hole, or PG 11 or PG 13.5
Trumpet Type Cable Connectors.

Supply (Input) and Output Pressure Gauges(E69F):
Mounted integral to converter.
Explosionproof Junction Box (E69F):

Assembled to Valve Actuator by Foxboro (E69F).

How to Order
1) SPECIFY MODEL NUMBER: E69F-		

____ __ – __Housing:

Bolted cover (weatherproof)................................................................................................ B
Threaded cover (explosionproof).........................................................................................T

Input Signal (a), (b) and Impedance:

4 to 20 mA dc (170 Ω).....................................................................................................................I
10 to 50 mA dc (27 Ω)....................................................................................................................H

Output Signal (c):
kPa

psi

20 to 100

3 to 1540 to 200

6 to 30................................................................................ 2................................................................................ 3—
3 to 27................................................................................ 3
7 to 125
1 to 18................................................................................ 7
7 to 220
1 to 32................................................................................ 8
Other ....................................................................................................................................................... X

Optional Features:

Explosionproof junction box.......................................................................................................................... J
Adjustable filter regulator without gauge.................................................................................................... R
Supply/output gauges..................................................................................................................................... S
Miniature junction box (1/2 in conduit connector) - front entry (d)..........................................................M
Miniature junction box (1/2 in conduit connector) - rear entry (d)............................................................ Pa) Converter action may be direct or reverse. (Specify.)b) Input signal may be split-ranged (i.e., 4 to 12 mA). (Specify.)c) When ordering valves not employing valve positioners, specify code 7 or 8 depending on the actuator spring range.
d) Cannot be used with Electrical Classification CS-E/LD-E.
HOW TO ORDER Continued on Next Page.

Page 86

This product and its components are protected by U.S. Patent 4,213,478.
Corresponding patents have been issued or are pending in other countries.

Converters (Current-to-Pneumatic)
E69, E69R

How to Order (continued)
1) SPECIFY MODEL NUMBER: E69R-				

__ – __

Housing:
Input Signal (a), (b) and Impedance:

4 to 20 mA dc (170 Ω)............................................................................................................................. I
10 to 50 mA dc (27 Ω)............................................................................................................................ H

Output Signal (c):
kPa

psi

20 to 100

3 to 1540 to 200

6 to 30......................................................................................... 2......................................................................................... 3—
3 to 27......................................................................................... 3
7 to 125
1 to 18......................................................................................... 7
7 to 220
1 to 32......................................................................................... 8
Other ................................................................................................................................................................ Xa) Converter action may be direct or reverse. (Specify.)b) Input signal may be split-ranged (i.e., 4 to 12 mA) (Specify.)c) When ordering valves not employing valve positioners, specify code 7 or 8 depending on the actuator spring range.

2) SPECIFY DIRECT OR REVERSE ACTION______________________________________________
3) SPECIFY OUTPUT SIGNAL___________________________________________________________
4) SPECIFY ELECTRICAL CLASSIFICATION______________________________________________
5) SPECIFY OPTIONAL FEATURES______________________________________________________
6) SPECIFY INFORMATION FOR INSTRUMENT TAG_____________________________________

Page 87

Converters (Current-to-Pneumatic)
IP24 Current to Pneumatic
Converter for Field & Service

IP24
Input
Signal range................................ 4 to 20 mA/0 to 20 mA/		
0 to 10 mA/0 to 10 V
Input resistance (at 20 °C)
Normal Version & Vers.
II 2 G EEx ia IIC T6 acc.
to ATEX.................................... <220 Ohms
Signal Range
0 to 10 mA / 0 to 10 V........... <1000 Ohms
OutputSignal range normal.................. 0.2 to 1 bar, 3 to 15 psi,		
20 to 100 kPa, 0.2 to 4 bar
Signal range normal.................. 1 to 0.2 bar, 15 to 3 psi,
		
100 to 20 kPa, 4 to 0.2 bar
Supply air..................................... 1.4 ±0.1 bar ( 20 ±1.4 psi)
Air consumption......................... <100 l/h
Supply air..................................... free of oil, dust, water		according to ISO 8573-1
Max. particle-size and -density.... Class 2
Max. oil contents.................... Class 3
Transmission performanceNon-linearity (terminal-based)... <0.3% of final value
Hysteresis..................................... <0.1%
Supply pressure dependence.... <0.25 %/0,1 bar
Temperature influence
Output normal........................ <0.3%/10 K
Output reverse....................... <0.5%/10 K
Sensitivety level.......................... <0.02%
Housing
Material..................................... Aluminum casting, finishedProtection Class....................... IP54 or IP65

			Pneumatic Signal Selection Relay

For conversion of a standard electrical signal
into a standard pneumatic signal.
For complete specification, refer to ProductSpecification Sheet PSS EVE0401 A-(en).
n High air capacityn Low input resistancen Easy adaptation of the Converter toranges 0 to 20 mA or 4 to 20 mA
n Mode of action normal or reversen Protection Class IP54 or IP65
n Version in II 2 G EEx ia IIC T6 acc. to ATEXn Stainless Steel 316 as option

How to Order___ ___ ___IP24

Input:

Signal Range 0 - 20 mA.................................................................................................................. -A
Signal Range 4 - 20 mA.................................................................................................................. -B
Signal Range 0 - 10 mA(a)............................................................................................................. -C
Signal Range 0 - 10 V(a)................................................................................................................. -D

Input:

Signal Range 0. -1 bar..............................................................................................................................K
Signal Range 3-15 psi...............................................................................................................................L
Signal Range 20-100 kPa........................................................................................................................ M
Signal Range 0.2-4 bar(a)........................................................................................................................ N
Signal Range 0.2-5 bar(a).........................................................................................................................P
Signal Range 0.2-6 bar(a)........................................................................................................................ Q

Mode of Action:

Normal (For Version ZZZ)................................................................................................................................ D
Normal (For Version EAA according to ATEX)..............................................................................................E		
Reverse...............................................................................................................................................................R

HOW TO ORDER Continued on Next Page.

Page 88

Converters (Current-to-Pneumatic)
IP24

How to Order (continued)					 ____ ___Electrical Certification:
II 2 G EEx ia IIC T6................................................................................................................................. EAA
Without....................................................................................................................................................ZZZ

Options:

Attachment Kit For Pipe Mounting................................................................................................................-A
Protection Class IP65....................................................................................................................................... -B
Calibration In kp/cm........................................................................................................................................-C
Assembled free of oil and Grease / Designed for Aux. Energy Oxygen................................................. -S

Tag No. Labeling:

Stamped With Weather Resistant Color.......................................................................................................-G
Stainless Steel Label Fixed With Wire........................................................................................................... -L

NOTES:(a) Only available with Mode of Action Code D

Special Version
Nr EP

Specifications

EP0194

Special setting: Input 4-20 mA ≥ Output 0.2-0.6 bar

EP0210Special setting: Input 4-20 mA ≥ Output 0.2-4 bar

EP0300IP24 in Stainless Steel Housing 316

EP0305Special setting: Input 4-20 mA ≥ Output 0.4-2 bar

EP0314

With 2 additional gauges (air supply, output)

Page 89

Control Valves (Actuators)
2A, P SERIES, 20F/P50, 30C/P110

2A Series Cylinder Actuatorwith Power Positioner

E69P Series Current-ToPneumatic Valve Positioners

The 2A Series Cylinders are used to actuate
dampers, gates, shutters, doors, and large
control valves. Power positioners are requiredwith all pneumatic cylinders for throttling action.
They convert the standard 20 to 100 kPa (3 to 15psi) controller signal to full supply pressure on
the cylinder piston and assure precise, positive
positioning. Cylinders are available in bore sizes
83 through 355 mm (3 1/4 through 14 in) withstrokes up to 810 mm (32 in).

The E69P Series Current-to-Pneumatic valve
Positioner converts a direct current milliampere
input signal into a pneumatic output signal
for positioning control valve actuators. It
eliminates the need for a separate converter
and a separate positioner. It is available with thefollowing input signal ranges (mA): 4 to 20, 4 to12, 12 to 20, 10 to 50, 10 to 30, 30 to 50 witha corresponding output sugnal of up to 240 or
420 dPa (35 or 60 psi, 2.4 or 4.2 bar of kg/cm2).

Models 20F/P50 & 30C/P110
Diaphragm Lever Actuators

P Series Pneumatic Spring
Diaphragm Actuators

The Models 20F/P50 and 30C/P110 Diaphragm
Lever Actuators are used to actuate dampers,
gates, and butterfly valves. The topworks ofthe size P50 and P110 Diaphragm Actuators
are made of vinyl­coated, die-cast aluminum.
Maximum input pressure is 420 kPa (60 psi).
Bracket material is cast iron, and the carbon
steel lever is predrilled. Lever strokes range
from 116 to 254 mm (4.56 to 10 in).

The P Series Actuators can operate a varietyof control valves, dampers, louvers, and gates.
They are lightweight field-reversible units withlow hysteresis and fast, precise response.
		
Model
P25
P50
P110

Maximum Stroke
mm
19
38
57

in
3/4
1 1/2
2 1/4

Effective Area
m2
0.016
0.032
0.071

in2
25
50
110

Material
Construction
aluminumwith castiron yoke

Nominal input signal: 20 to 100 or 40 to 200 kPa(3to 15 or 6 to 30 psi).

Page 90

Control Valves (Positioners)
SRP981, SRI986, SRI983, SMP981

SRP981 Pneumatic Positioner

SRI986 Electro-Pneumatic
Positioner

The SRP981 Positioner is designed to control
pneumatic valve actuators with pneumatic control
signals. It offers an easy adjustment by two mechanicalscrews.
• Split range up to 4-fold possible• Input Signal from 0.2 … 1 bar (3 … 15 psig)• Supply pressure up to 6 bar (90 psig)• Single or double acting• ATEX Explosion protection II2 G EEx c(constructive safety)• Stainless steel 316 housing as option

The SRI986 Positioner is designed to control
pneumatic valve actuators from control systems andelectrical controllers with electric control signals. It
offers an easy adjustment by two mechanical screws.
• Split range up to 3-fold possible• Input Signal from 0 to 20 mA or 4 to 20 mA• Explosion protection:o II 2 G EEx i (intrinsic safety) according to ATEXo Intrinsic safety according to FM and CSA• Ambient temperature –40 to 80°C (–40 to 176°F)

SRI983 Electro-Pneumatic
Positioner

SMP981 Pneumatic Position
Transmitter

The SRI983 Positioner is designed to control
pneumatic valve actuators from control systems
and electrical controllers with electric control
signals. It offers an easy adjustment by two
mechanical screws.
- Explosion protection:• II 2 G EEx d (flameproof) according to ATEX
explosion proof according to FM and CSA• Ambient temperature –40 to 80°C (–40 to
176°F)For the conversion of linear or rotary
movements actuators into a 0.2 to 1 bar
pneumatic signal.
• Universal matching to all strokes by means ofdiffering range springs.
• Simple installation and adjustment. Zero andrange settings non-interactive• ATEX Explosion protection II2 G EEx c(constructive safety)

For more information, please refer to our main
instrument catalog, Foxboro Measurements andInstruments Catalog (PN FX-0181).
Page 91

Accessories and Supplies
Humitex Circular and Strip Charts
Foxboro recording charts complement the precision and fine craftsmanship of our wide range ofrecording instruments. Charts combine the essential elements of special high-quality paper with rigidly
controlled printing and trimming procedures. Over 17,000 different charts can be supplied in ranges
and calibrations to meet the requirements of practically any process recording application. Special
charts and new ranges not listed in the Chart Catalog and Dial List (Catalog 600) may require a newelectro printing plate for specific customer needs. Contact our nearest Branch Office for Catalog 600.

Chart Purchase Agreement
Any Foxboro instrument user with an estimated yearly requirement of 500 boxes of circular and/or
500 boxes of strip charts or more may request a Chart Purchase Agreement. Under the agreement
system, quantity prices are applied to the individual chart purchase order. The agreement is
continuous, reviewed yearly for actual purchases, and a 30-day advance notice to changes is given.
Refer to Foxboro with your annual requirements to receive a Chart Purchase Agreement.
Minimum Order Quantity—15 boxes of circular charts or 15 boxes of strip charts.
Toll Free Number
To order charts, call 888-FOXBORO (888-369-2676).
Types of Calibration Charts Available for Circular and Strip Charts
Uniform, Differential Pressure, Flow (Type 27), Vacuum Pressure, Square Root, Vapor Pressure, Dairy
Charts, Dew Point, Thermocouple, Thermopile, Flow-Open Channel Weir and Flume, Log Function,
Specific Gravity, and Balsbaugh.

Circular Chart Specifications
Standard circular charts are printed with grayink, 1 side only, on 0.122 mm (0.0048 in) thick
paper. Circular charts with heavy paper are
printed on 0.178 mm (0.0070 in) paper. Charts
are packaged 100 to a box.
Instrument
Chart Size
125 mm (5 in) Side-Mounted
Recorder…………………………125 mm (5 in) (a)12 Series Recorders………………300 mm (12 in)
40/740 Series Recorders ………..300 mm (12 in)
39A Series Flow Recorders…..300 mm (12 in) (a)39 B Series Flow Recorders…………...MW Series

300mm (12in)(a)740R Series Recorders…………...300 mm (12 in)a) Always printed on heavy paper.

Foxboro also prints circular charts for
American Meter, Arcco-Anubis, Bailey,
Bristol, Gotham (Ametek), Honeywell (Brown),
Mercury, Palmer, Ranarex (Permutit), Rockwell,
Taylor, UGC Industries, Warren Controls(Tagliabue), and Weksler which are made totheir specifications.

Page 92

Options for 300 mm (12 in) Circular Charts:
Description

Code (b)Heavy Paper		
Backprinting (c)		
Mullins Slotting (d) 		
Overprinting of Two-Color Charts (f)		
Color of Ink on Chart:
		 Green				 Scanner Blue Ink 				 Black 				 Aqua				 Dark Blue 		
Shrink Wrap 		

-HP-BP-MS-HP-OP-GRN-BLU-SBI
-BLK
-AQUA
-DBLU
-SW

b) Add as suffix to circular chart number.
c) Add Foxboro backprint number.
d) Specify drop time and/or day of the week.
e) Specify time.
f) Must be available as an existing chart; see Catalog 600.

Accessories and Supplies
Strip Chart Specifications
4 in Roll,100 mm Roll, and SCAN-FOLD charts can be supplied with or without time mark numbering.
Standard time mark numbering is the hour of the day printed on the left side of the chart at 2-hourintervals (on the even hour). The standard speed for the 4 in Roll and SCAN-FOLD is 3/4 in/h whichcoincides with the chart drive speed. The standard speed for the 100 mm Roll is 20 mm/h.
A large selection of non-standard time mark numbering is available to match the chart speeds of yourrecorders. Refer to Catalog 600 for a listing.
4 in Roll Chart............................................................................................................ One 30-day chart per box
100 mm Roll Chart.................................................................................................... One 30-day chart per box
SCAN-FOLD Chart................................................................................................... Two 16-day charts per box

Recorders

Chart

E20S
53
54
64
120
220S

4 in Roll or SCAN-FOLD
For 4 in Roll, specify chart number
suffix -T if time numbering is required.

E27R
126S
127S
226S
227S
760R

For SCAN-FOLD, specify chart number
suffix -6TX if time numbering is required.
100 mm Roll only
Specify chart number suffix -T if time numbering is required.

761R			L0120NP
		
		
762R
		
		
751R
		

Fold -

L0122RQ		50 Divisions
L0122RS		0-100 Linear

Roll -

L0122RR		50 Divisions
L0122RT		0-100 Linear

Fold Roll -L0122RS		0-100 Linear
L0122RT		0-100 Linear

Page 93

Accessories and Supplies
Pens and Pen Armsn Large, Disposable, Fiber Tip Pens for 		

Circular Chart Recording Instruments

n Designed to Write 365 meters 			(1200 feet)n Prepackaged 6 Pens per Card

Recording40, 40M, 40P, and 39 Series Rectangular Case Recorders:				
Pen Location
1-Pen
2-Pen
		inner		outer3-Pen
		inner
		center		outer4-Pen
		inner		 next to inner
		 next to outer		outerPen Arm, Item 1

Fiber Tip Pens (Package of 6), Item 2Part		Identification		
Number		 Holes
Color
0044897
2
Red

A Length
Part			Number		 mm		 inL0121CH12
0.47

M0122AC0044897

1
2

VioletRed

L0121CM
L0121CH

6
12

0.24
0.47

M0122AC0044897
M0122AB1
2
3

VioletRed
Green

L0121CM
L0121CH
L0121CU

6
12
18

0.24
0.47
0.71

M0122AC0044897
M0122AB
M0122AA12
3
0

Violet
Red
Green
Blue

L0121CM
L0121CH
L0121CU
L0121DA

6
12
18
24

0.24
0.47
0.71
0.94

12R Series Round Case Recorders:		

Pen Arm, Item 1

Fiber Tip Pens (Package of 6), Item 2		
Pen Location

Part		Identification		
Number		 HolesColor

A Length
Part			Number		 mm		 in

1-Pen
2-Pen
		outer		 inner

0044899

2

Red

L0121CH12

0.4700469650044899

3
2

VioletRed

L0121CR
L0121C J

6
12

0.24
0.47

12RD (1 to 3 Pen) and 12RM (1 and 2 Pen) Series Round Case Recorders:		

Pen Arm, Item 1

Fiber Tip Pens (Package of 6), Item 2		
Pen Location

Part		Identification		
Number		 HolesColor

A Length
Part			Number		 mm		 in

1-Pen
2-Pen
		inner		outer3-Pen
		inner
		center		outer

0044899

2

Red

L0121CT18

0.7100469640044899

1
2

VioletRed

L0121CR
L0121CT

12
18

0.47
0.71

M0122NE00469640044899012VioletRed
Green

L0121CR
L0121CT
L0121CZ

12
18
24

0.47
0.71
0.94

To order, call 888-FOXBORO (888-369-2676).

Page 94

Accessories and Supplies
Recording Controllers
40, 40M, and 40P Series Rectangular Case Recording Controllers:Pen Arm/lndex, Item 1
Fiber Tip Pens (Package of 6), Item 2						A Length
Controller
PenPart
Identification
Part
		
			
Type
Location
NumberColor
Numbermm		 in				 Holes or Color
1-Pen00448972
Single Action
		Index
0032301
Plain
Red
L0121CH
12
0.47
		
2-Pen
Single Action		innerwith		outer
Additional Pen
Index

M0122AC0044897
M0122LT

1
2
RedVioletRed

L0121CR
L0121CT

12
18

0.47
0.71

		
3-Pen
Single Action 		 innerwith 2 		 center
Additional Pens 		 outer
		
Index

M0122AC0044897
M0122AB
M0122LT

1
2
3
Red

VioletRed
Green

L0121CR
L0121CT
L0121CZ

12
18
24

0.47
0.71
0.94

		
1-Pen
		
Index
Triple Setting 		 inner			center			outer

00448972

M0122LS
M0122LT
M0122MA

Violet
Red
Green

Red

L0121CY

24

0.94

		
2-Pen
		innerRatio
			outer		
Index

M0122AC0044897
M0122LS

1
2
VioletVioletRed

L0121CR
L0121CT

12
18

0.47
0.71

		
3-Pen
		inner
Ratio with inner		 center
Additional Pen
			outer		
Index

M0122AC0044897
M0122AB
M0122LS

1
2
3VioletVioletRed
Green

L0121CR
L0121CT
L0121CZ

12
18
24

0.47
0.71
0.94

Red

L0121CT

18

0.71

Red
Violet

L0121CT
L0121DB

18
24

0.71
0.94

		

1-Pen
00448972
Index			Duplex			inner
M0122LTRed			outer
M0122LS
Violet		
2-Pen
Auto-Selector, 		 inner
M0122AC
1
Dual, or Duplex 		 outer
00448972
withIndex			Additional Pen 		 inner
M0122LTRed			outer
M0122LS
Violet1.	On multiple pen recorders, if pens L0121CR (Violet), LC 121CS (Black), L0121CP (Green), or L012 CQ (Blue) are used as arcing pens, then the
red inner pen (L0121CJ) must be used. Its length is 6 mm (0.24 in)
2. Black pens may be substituted for red or violet as follows:
Use L0121CN for pen arm lengths of 6 mm (0.24 in)
Use L0121CS for pen arm lengths of 12 mm (0.47 in)
3. If replacing box pens with fiber tip pens, the color dot on the pen arm (Item 1 in drawing on preceding page) must be removed.

Disposable Fiber Tip Pen Cartridges for 40P/40M/12 and 740 Chart Recorders:		A-Lengthmm		 in
6
12
18
24

0.24
0.47
0.71
0.94

Red

Violet

Green

Blue

Black

L0121CJ
L0121CH
L0121CT
L0121CY

L0121CM
L0121CR
L0121CW
L0121DB

L0121CK
L0121CP
L0121CU
L0121CZ

L0121CL
L0121CQ
L0121CV
L0121DA

L0121CN
L0121CS
L0121CX
L0121DC

Page 95

Trademarks

Alumel is a trademark of Hoskins Manufacturing Company.
Aminco is a trademark of American Instrument Company.
Bakelite is a trademark of Union Carbide Corporation.
Carpenter is a trademark of Carpenter Technology Corporation.
Chromel is a trademark of Hoskins Manufacturing Company.
Dacron is a trademark of E. I. duPont de Nemours & Company, Inc.
Duranickel is a trademark of Huntington Alloys, Incorporated.
Fluorinert is a trademark of 3M Company.
Fluorolube is a trademark of Hooker Chemical Corporation.
Grafoil is a trademark of Union Carbide Corporation.
Hastelloy is a trademark of Stellite Division of Cabot Corporation.
HART ia a trademark of the HART Foundation
Inconel is a trademark of Huntington Alloys, Incorporated.
Kel-F is a trademark of 3M Company.
Kynar is a trademark of The Pennwalt Corporation.
Linatex is a trademark of Wilkinson Process Rubber, Ltd.
Monel is a trademark of Huntington Alloys, Incorporated.
Ni-Span is a trademark of Huntington Alloys, Incorporated.
Noryl is a trademark of The General Electric Company.
Ryton is a trademark of Phillips Petroleum Company.
Teflon is a trademark of E. I. duPont de Nemours & Company, Inc.
Tefzel is a trademark of E. I. duPont de Nemours & Company, Inc.
TriClamp is a trademark of Ladish Company.
Viton is a trademark of DuPont Dow Elastomers L.L.C.
17-4 PH is a trademark of Armco Steel Corporation.
17-7 PH is a trademark of Armco Steel Corporation.
The following are trademarks of The Foxboro Company:
CONSOTROLDEWCEL
d/p Cell
DYNATHERM
FOXBOROHUMITEXSCANFOLD
The lntelligent Automation People is a Service Mark of The Foxboro Company.

Page 96Product Index

101 Series Shelf.................................Page 83
120 Series Recorders................................. 83
130 M-N Gap Controller............................ 83
138 Series.................................................... 83
2A Actuator................................................. 90
E69P Series.................................................. 90
Absolute Pressure Transmitters................ 15
Actuators...................................................... 90
Adjustable Range Transmitters....................3
Air Filters...................................................... 32
Air Regulators.............................................. 32
Airsets........................................................... 32
Alarm Actuators.......................................... 83
Alarm Lamps............................................... 83
Alarms.......................................................... 83
Analog Computers..................................... 79
Auto Selector Control Systems................. 69
Bias Stations................................................ 76
Buoyancy Transmitter................................ 32
Bypass Manifolds........................................ 32
Chart Pen Arms........................................... 94
Chart Pens................................................... 94
Chart Purchase Agreement....................... 92
Computers................................................... 79
CONSOTROL Shelf..... 66, 67, 69, 71, 73, 83
CONSOTROL............................................... 83
Control Stations.......................................... 83
Control Valves............................................. 90
Controller.........................................42, 45, 83
Converters................................................... 84
Current to Pneumatic Converter.............. 85
d/p Cell.................................................... 3, 20
Density Transmitter.................................... 20
Dew Point Measurement System.............. 40
DEWCEL....................................................... 40
Diaphragm Differential Pressure Element.... 56
Diaphragm Pressure Seals........................ 39
Electronic Servo Elements......................... 62
Filled Thermal System Elements.............. 59

Flanged Transmitters................................. 20
Flanged Unions........................................... 34
Flow Integrators.............................................9
Gap Controllers.......................................... 83
Gauge Pressure Transmitters.................... 12
Humitex Circular Charts............................. 92
Humitex Strip Charts.................................. 92
IFOA Series.................................................. 36
Indicating Controllers............. 45, 67, 69, 71
Indicating Transmitters.............................. 32
Indicators ........................��������� 51, 67, 69, 71
Integral Flow Orifice Assemblies.............. 36
Integrators......................................................9
Lamps........................................................... 83
Lever Actuators........................................... 90
Liquid Level Transmitters.....................20, 32
Loading Stations......................................... 76
Low Range Transmitters...............................3
Manual Station............................................ 76
Model 101................................................... 83
Model 102................................................... 83
Model 110................................................... 71
Model 11AH................................................ 15
Model 11AL................................................. 15
Model 11AM................................................ 15
Model 11DM..................................................7
Model 11GH................................................ 12
Model 11GM............................................... 12
Model 120................................................... 83
Model 127S................................................. 73
Model 12A................................................... 28
Model 12AS................................................. 28
Model 130M-N............................................ 83
Model 130F................................................. 67
Model 130FD.............................................. 83
Model 130M................................................ 67
Model 130MD............................................. 83
Model 130P................................................. 67
Model 130PD...............................................83
Continued on next page

Page 97

Product Index continued

Model 135.................................................. 76
Model 138.................................................. 83
Model 139PP.............................................. 18
Model 13A.................................................... 3
Model 13DEMP.......................................... 23
Model 13DMP............................................ 23
Model 13F.................................................. 20
Model 13FE................................................ 20
Model 13H.................................................... 3
Model 14A.................................................... 9
Model 15A.................................................... 3
Model 15F.................................................. 20
Model 167LP.............................................. 26
Model 17BS................................................ 32
Model 17BT................................................ 32
Model 20F.................................................. 90
Model 30C.................................................. 90
Model 40M...........................................42, 51
Model 40P............................................42, 51
Model 43AP................................................ 45
Model 43APG............................................. 54
Model 45P.................................................. 32
Model 556 / 557........................................ 79
Model E69F................................................ 85
Model E69R................................................ 85
Model FRS02.............................................. 31
Model FRS03.............................................. 31
Model IP24................................................. 84
Model P110................................................ 90
Model P417................................................ 48
Model P50.................................................. 90
Model PC191............................................. 81
Model PM960............................................. 78
Model PS953.............................................. 82
MR Series Receiver Gauges..................... 65
MR Series.................................................... 65
Orifice Plates.............................................. 34
Pneumatic Flow Integrator......................... 9
Pneumatic Transmitters....................3, 7, 12Page 98

Positioners.................................................. 90
Pressure Measuring Elements.................. 57
Pressure Seals............................................ 39
Pressure Transmitter Seals....................... 23
Receiver Gauges........................................ 65
Recorders.......................... 51, 54, 73, 83, 94
Recording Indicating Controllers......42, 45
Remote Mounted Controllers.................. 83
Servo Elements.......................................... 62
Spring Diaphragm Actuators................... 90
Switching Stations...................................... 76
Temperature Transmitters........................ 28
Thermal Elements...................................... 59
Thermowells............................................... 40
Transmitters.............................................. 3, 7
Type 37....................................................... 56
Type 70....................................................... 65
Unions......................................................... 34

Page 99

Page 100

OrderingSpecifying Your Instrument or Accessory
You can easily specify many instruments and accessories described in thiscatalog. Sections covering our most popular items include all the technical data
you need to know for most applications. To specify the appropriate item, simply
follow the step-by-step procedure at the end of each description. Your Foxboro
representative can help also.
For Example:
Choose the product code.
Itemize the information needed to complete the specification.
Ordering:
To order, contact your Foxboro representative.
Or call, in North America: 1-866-746-6477, or International: 001-508-549-2424.
Or visit www.buyautomation.com/officelocator.
Give us the ordering information using this catalog. Then tell us where you want
the order shipped and when you need it delivered. We’ll do the rest.
Need Help?
If you have questions, let us know. We’re ready, and eager, to assist you.
Applications are a Foxboro specialty. We can help you figure out the best
instrument for your purpose.
Instrument catalog available — ask your Foxboro representative about it.

1) SPECIFY MODEL NUMBER: 13
Transmitter:

–

–

–

316 ss body.......................................................................................................... A
Monel body.......................................................................................................... A4
Hastelloy C body (supplied without process connectors)............................. A5

Span Limits:

5 and 62 kPa or 20 and 250 inH2O ΔP....................................................................... M
50 and 210 kPa or 200 and 850 inH2O ΔP................................................................ H

Body Material:

316 ss (13A only)..................................................................................................................... S
Monel (13A4 only)................................................................................................................... M
Hastelloy C (13A5 only) (Process Connectors code 0 below must be specified........... C

Process Connectors:

1/4 NPT..............................................................................................................................................
1/2 NPT..............................................................................................................................................
R1/4....................................................................................................................................................
R1/2....................................................................................................................................................
Welding neck for 14 x 21 mm tube and 1/2 in Schedule 80 pipe............................................
None, body tapped for 1/4 NPT....................................................................................................

1
2
3
4
6
0

Optional Features: (also see previous page)

Zero elevation kit........................................................................................................................................ L
Zero suppression kit.................................................................................................................................. R

2) SPECIFY OUTPUT SIGNAL___________________________________________________
3) SPECIFY CALIBRATED DIFFERENTIAL PRESSURE RANGE_______________________
4) SPECIFY ADDITIONAL OPTIONAL FEATURES_________________________________
5) SPECIFY INFORMATION FOR INSTRUMENT TAG______________________________

33 Commercial Street
Foxboro, Massachusetts 02035-2099 U.S.A.

Your Foxboro representative:Pneumatic Instruments Catalog

Invensys plc
3rd Floor
40 Grosvenor Place
LondonSW1X 7AW
United Kingdom
Tel: +44 (0)203 155 1200
www.invensys.com

Foxboro33 Commercial Street
Foxboro, Massachusetts 02035-2099 U.S.A.
+1(866) 746-6477 Toll-free within U.S.A+1(508) 549-2424 Global

Foxboro Eckardt
Pragstrasse 82
D-70376 Stuttgart
GermanyTel +49 (0)711 502-0
Fax +49 (0)711 502-597
Email: salessupport@foxboro-eckardt.deInvensys, the Invensys logo, ArchestrA, Avantis, Eurotherm, Foxboro, IMServ, InFusion, SimSci-Esscor, Skelta, Triconex, and Wonderware are trademarks of Invensys plc, its subsidiaries oraffiliates. All other brands and product names may be the trademarks or service marks of their representative owners.
© 2013 Invensys Systems, Inc. All rights reserved. No part of the material protected by this copyright may be reproduced or utilized in any form or by any means, electronic or mechanical,
including photocopying, recording, broadcasting, or by any information storage and retrieval system, without permission in writing from Invensys Systems, Inc.
PN FX-0180

Rev. 04/13