Dentaport ZX – Reliable Apex Location – Perfect Endodontics
Curing with High-Powered Parallel Precision

Thinking ahead. Focused on life.

Thinking ahead.

Focused on life.

Key product features include the apex locator and optional canal-preparation and lightcuring functions. In addition, the multi-joint
feature lets you change the function just bychanging the handpiece.

4

Main Features
Modular Root-Canal Treatment Instrument Dentaport ZX
By attaching modules to the Root ZX Module as an apex locator,
the dentist can prepare canals and use a LED curing light device.
Multi-Joint Tube
Simply switch handpieces to use whichever function you need.
Compact and Light Handpieces
Both the motor and the handpieces of curing light are ergonomicallydesigned to be well-balanced and fit perfectly in your hand.
Programs
The instrument’s programs can store three sets of values
simultaneously.

Dentaport ZX – Highlights at a Glance

Apex Locator
Accurate Measurement
The accuracy of the measurement is not affected by the presenceor absence of blood, other discharges, electrolytes, saline, tapwater or hydrogen peroxide.
No Zero-Adjustment
It is not necessary to set the device on zero before measuring
each individual canal.
Automatic Calibration
Automatic calibration ensures accuracy and eliminates the effectof changes in temperature, moisture, etc. inside the canal evenduring the treatment.

Canal Preparation
Prepare Canals while Simultaneously Taking Measurements
Accurate and precise preparation of the root canal.
Flexibility: 8 Speed Ranges
150, 200, 250, 300, 400, 500, 600, 800 rpm
Five-Function Auto Control System
Auto Apical Reverse
Auto Torque Reverse
Auto Start and Stop
Auto Apical Slowdown
Auto Torque Slowdown

LED Curing LightPowerful Parallel Light Beam, with Intensity 1,000 mW/cm2
Maintains power even at a distance of 10 mm.
Compact Design
It easily reaches even the distal side of molar 7.

5

Industry-Standard Apex Locator
Root ZX Module

Root ZX Module
The accuracy of the measurement is notaffected by the presence or absence of blood,other discharges or electrolytes. Fully automatic, it is not necessary to set the device on
zero before measuring each individual canal.
Automatic calibration ensures accuracy andeliminates the effect of changes in temperature, moisture, etc. inside the canal evenduring the treatment. Both a meter and an
audible signal tell you the position of theapical constriction.

Accurate Measurement
The ratio technique used in the instrument provides a highly
accurate detection of the file’s location. The accuracy of themeasurement is not affected by the presence or absenceof blood, other discharges, electrolytes, saline, tap water orhydrogen peroxide.

2Large color display

2

Colored bar indicator

3

Changeable
working length dot

4Green indicator for quick
recognition of the apex

3

No Zero-Adjustment
It is not necessary to set the device on zero before measuring
each individual canal.

114

Automatic Calibration
Automatic calibration ensures accuracy and eliminates the effectof changes in temperature, moisture, etc. inside the canal, evenduring the treatment.
Large Color Display
The large color display is easy to read, providing both precisionand high contrast. The action of the meter in the display corresponds exactly to the tactile sensation of using the file.

6

Changeable
Working-Length Dot
The working-length dot is easy
to change. It can be changed
between 2 and APEX.

Canal-Preparation Instrument
By showing the file position and using 5 automatic control
functions, this instrument allows you to prepare canals
with accuracy and precision up to any desired point inside
the canal – and reduces file jamming and breakage.

5 Automatic Control Functions

Accurate preparation
Auto Apical Reverse
Foot Switch for Easy Operation
You can attach the optional foot switch to use it
as the micro motor on the dental units. When aroot-canal measurement isn’t possible, you canactivate the motor by the foot switch. The footswitch is especially helpful for preparing theupper part of the root canal.

File stops and reverses when
it reaches the end of the set
working length. You can also
set stops manually or adjust
the interval between stopping
and reversing.

7

Dentaport_ZX_Multi_en.indd 7

15.03.2007 16:49:21 UhrEasily Add and Change Functions –both Preparation and Light-Cure Functions are Available

12.5mmFor attaching the canal-preparation and curing light module
to the instrument, you can use a connection point for the handpieces that provide these functions. To change the function,
simply exchange the handpiece.

165 mm (1:1)

A Smaller, Lighter, Finer Motor Handpiece
Ergonomically Designed to Minimize Fingertip Stress

Comfortable Handling

Weight: only 70 g (approx.)

The handpiece is both
comfortable and sensitive.

Compact head size:
only 12.5 mm
Speed: 150 to 800 rpm
with 8 speed ranges

8

Reduced file breakage
Auto Torque Reverse

No switch to turn on and off
Auto Start and Stop

Precise control
Auto Apical Slowdown

Reduced file breakage
Auto Torque Slowdown

File stops and reverses if
the torque goes over the set
torque limit. Torque limit canbe set at 11 levels.

Motor starts when file isinserted and stops when it iswithdrawn.

File slows down as it approaches the end of the set working
length, ensuring safe and surepreparations.

File slows down as the torqueapproaches the set torque limit.

Parallel-Light Beam isPowerful even at a Distance

In Both Cases, You Will Haveenough Adhesion Strength

Even at a distance of 10 mm,
the light retains 80% of its
power.

It can be used for the bondingof the fiber post because of
the high power at a distanceof 10 mm.

Easily reaches the distal sideof molar 7, normally a difficult
task for instruments using alight guide.

10 mm
1 mm

There is no difference of the size of the circle spot between
10 and 1 mm. So, the power does not change even at thedistance of 10 mm.

High-Powered LED Curing Light
Set Irradiation Time
Set timer from 5 to 40 seconds
in 5 second steps

300°

16 mm

A powerful curing light – with intensity 1,000 mW/cm2 –that works even at a distance of 10 mm. It delivers an evenintensity both in the center and at the edges of the irradiation field. Composite curing time: 20, 10 sec., bonding time:
10 sec.

175 mm (1:1)
Light, Compact Handpiece
Ergonomic design for optimum
performance and handling.

Compact Head Rotates
The light and compact headalso rotates for ideal handlinginside the oral cavity. Easily
reaches molar areas.

Perfectly Balanced
Head diameter: 15.5 mm
Head height: 16.0 mm
Weight: approx. 60 g
Head rotation: 300°

Ergonomically designed for
optimum handling. The handpiece is perfectly balanced anddesigned to fit the hand exactly, so it reduces the stressand load on your fingertips.
92Specifications
12Root ZX Module
Rating: DC 4.5 V
Power Comsumption: 0.2 VA
Size: W 105 X D 105 X H 115 mm
Weight: approx. 400 g
Canal-Preparation
and Curing Light Module
Power consumptions: 20 VA
Charger: Rapid Charge TypeSize: W 105 X D 127 X H 115 mm
Weight: approx. 430 g

1345

Main Tubing
For Motor Handpieceand Light Cure
Motor Handpieceand Contra Angle
Diameter: max. 18 mm
Length: 165 mm
Weight: approx. 70 g(including contra head)

34

Light Curing Handpiece
Diameter: max. 18 mm
Length: 150 mm
Weight: approx. 60 g

5Trade name: Dentaport ZX
Model:DP-ZX
Manufacturer: J. Morita Mfg. Corp.
Apex Locator

Accessories
616Root ZX Module Accessories
1 x probe cord
3 x file holders
5 x contrary electrodes
1 x canal-length tester
3 x DC 1.5 VAA battery

Apex Locator and Canal-Preparation Handpiece
Accessories
1 x handpiece rest
1 x AR spray
1 x AC adapter(100 V–240 V 50/60 Hz)

12346

12356

Apex Locator and Light Cure
Accessories
1 x handpiece rest
1 x disposable covers(1 box of 100)
1 x AC adapter(100 V–240 V 50/60 Hz)

Apex Locator, Canal-Preparation Handpiece and Light CureAccessories
1 x handpiece rest
1 x AR spray1 x disposable covers(1 box of 100)
1 x AC adapter(100 V–240 V 50/60 Hz)

Design and specifications are subject to change without notification.

10

1

2

3

4

5

6

Optional

Foot Switch

Motor Tri Auto ZX
Cordless Canal-Preparation and Measurement Instrument
This instrument is a cordless handpiece that measures andlocates the position of the file tip during preparation for safer
and quicker treatment. When used with nickel-titanium files,
which have exceptional flexibility, it can prepare even curved
canals – which are usually quite difficult to treat.
Three Auto Safety Functions
Auto Start and Stop
Auto Torque Reverse
Auto Apical Reverse
Canal-Measurement Capability
Easy Handling
File tip-location identified by LEDs on handpiece and byaudible signal.
Battery-Powered Motor eliminates the bother of a cord.

Manufacturer: J. Morita Mfg. Corp.

Related Products
EndoWave nickel-titanium
files meet perfectly all therequirements for root-canal
preparation with a handpiece. They are optimallydesigned for use with theDentaport ZX. Only five
files are required for preparation in a straightforward
sequence.

The triangular cross section of the fileminimizes the risk of fracture

Anti-Screwing DesignUnique Safety Tip
Optimum Cutting Edges
Extremely SmoothSurface Texture

Manufacturer: J. Morita Europe GmbHFor detailed information, please see separate EndoWave leaflet.

Color-coded rings ensure the correct
file can be quickly selected

The EndoWave file system includes
two basic procedure kits:
Kit A for normal root canals, Kit B for
severely curved and narrow root canals

11Thinking ahead. Focused on life.

Treatment Units
Instruments
Laser Equipment
Laboratory Devices
Educational and Training Systems
Auxiliaries

Developed and Manufactured by
J. MORITA Mfg. Corp.
680 Higashihama Minami-cho, Fushimi-ku, Kyoto,
612-8533 Japan
Tel: +81-75-611-2141, Fax: +81-75-622-4595
http://www.jmorita-mfg.com

Distributed by
J. MORITA CORPORATION
33-18, 3-Chome, Tarumi-cho Suita City, Osaka, 564-8650 Japan
Tel: +81-6-6380-2525, Fax: +81-6-6380-0585
http://www.asia.morita.com http://www.oceania.morita.com
J. MORITA USA, Inc.
9 Mason lrvine, CA 92618 U.S.A.
Tel: +1-949-581-9600, Fax: +1-949-465-1095, http://www.jmoritausa.com
J. MORITA EUROPE GMBH
Justus-von-Liebig-Strasse 27A, D-63128 Dietzenbach, Germany
Tel: +49-6074-836-0, Fax: +49-6074-836-299, http://www.jmoritaeurope.com
Siamdent Co., Ltd.
2991/42 (JMTI Plaza), Visuthanee Hi-Tech Office Park, Ladprao Road,Klongchan, Bangkapi, Bangkok 10240, Thailand
Tel: +66-2-370-1346, Fax: +66-2-370-1345

418700/2552

In 1916, Junichi Morita started toimport products of the leadingdental equipment manufacturers
into Japan, where demands formodern dentistry were growing.
His venturesome attempts ofsupplying selected products fororal healthcare has grown steadily
by receiving valuable support andguidance from the dental profession.
His enterprising spirit lives throughthe decades, and all Morita Group
Companies join in continuing topursue marketing, distribution andservices, as well as R&D andmanufacturing, in collaboration withworld leaders in healthcare products
and research organizations.

Diagnostic/Imaging Equipment