mU

B nd P u

EN

Description:
Ultra-Blend™ plus is a light-cured, radiopaque, dentin-adhering liner and basematerial containing calcium hydroxide and calcium hydroxyapatite in a urethane
dimethacrylate (UDMA) base. Ultra-Blend plus has a high filler load to minimize
shrinkage and a higher viscosity to facilitate ease of placement. It can be easily
contoured with a high- or low-speed handpiece and is available in dentin andopaque white shades.
Indications:
UltraBlend plus is self-adhering and used as a cavity liner and/or base material
prior to placement of restorative materials.
Directions for Use:
A. General
1. Attach Black Micro™ tip securely onto syringe. For larger preparations, use
the Black Mini™ tip.
2. Prepare area with a quality bonding system according to manufacturer’s
instruction.
3. Apply a .5-2mm layer of Ultra-Blend plus to the desired area.
NOTE When applying to an area with difficult access, express a small
amount to form a ball at the tip. Use the tip as a ball applicator to prevent
over-dispensing.
4. Light cure with VALO™ LED Curing Light on Standard Power for 10 seconds(for lights with output <600mW/cm2 cure 20 seconds or for lights with
output >600mw/cm2 cure for 10 seconds).
NOTE For base build up, start with a 0.5mm layer and light cure. Follow with
2mm increments, curing between each increment.
5. Ultra-Blend plus will bond to surfaces prepared with dentin bonding
agents and to dry dentin.
B. Dentin Liner
1. Rinse exposed dentin with water and air dry.
2. Apply a quality bonding system according to manufacturer’s instruction.
3. Place Ultra-Blend plus precisely and in exact volume desired with a Black
Micro tip.
4. Light cure with VALO™ LED Curing Light on Standard Power for 10 seconds(for lights with output <600mW/cm2 cure 20 seconds or for lights with
output >600mw/cm2 cure for 10 seconds).
C. Protective Base
1. Rinse exposed dentin with water and air dry.
2. Apply a quality bonding system according to manufacturer’s instruction.
3. Attach a Black Micro tip to syringe if base area is small; use a Black Mini
tip for larger areas.
4. Apply Ultra-Blend plus and light cure with VALO™ LED Curing Light onStandard Power for 10 seconds (for lights with output <600mW/cm2 cure
20 seconds or for lights with output >600mw/cm2 cure for 10 seconds).
5. If final base is to be thicker than 2mm, apply 0.5mm (or less) as the firstlayer and light cure 10 seconds. Add additional layers of 2mm thicknesses
as needed.
Note: Ultra-Blend plus light cures to a depth of 3mm.
D. Pulp Capping
NOTE: Use Ultra-Blend plus for near pulp (pink) and small mechanical nonhyperemic exposure. If larger exposure and/or if hyperemic pulp, endodontic
therapy should be initiated.
1. Apply Consepsis™ antibacterial solution with plastic Mini Dento-Infusor™ tip
for 60 seconds using a gentle scrubbing action. Do not rinse. Gently air dry.
2. Apply Ultra-Blend plus with the Black Micro tip to the dry dentin at theborder of the exposure then over the exposure or over the thin pink dentinof the near exposure.
NOTE: Minimize covering dentin to maximize area available for dentine
bonding techniques.
3. Light cure with VALO™ LED Curing Light on Standard Power for 10 seconds(for lights with output <600mW/cm2 cure 20 seconds or for lights with
output >600mw/cm2 cure for 10 seconds).
4. Apply a quality bonding system according to manufacturer’s instruction
and restore with product of choice.
E. Opaquer
1. Roughen metal with bur or microabrasion.
2. Rinse exposed metal to be opaqued with water and air dry.
3. Apply a quality bonding system according to manufacturer’s instruction
and light cure.
4. Apply Ultra-Blend plus over metal incrementally with no layer thicker than
2mm. Light cure with VALO™ LED Curing Light on Standard Power for 10
seconds (for lights with output <600mW/cm2 cure 20 seconds or for lights
with output >600mw/cm2 cure for 10 seconds).
5. Restore with composite or product of choice.
Precautions:
WARNING: Chemical may set in tip after a few hours. To avoid clogging, remove
tip and re-cap with Luer Lock Caps between each use. Always verify material flow
prior to applying intraorally. If resistance is met, replace tip and re-check. Use
only recommended tips.
1. Carefully read and understand all instructions and SDS before using UltraBlend plus. Use only as directed.
2. Clinician and patient should wear UV protective eyewear when curing
Ultra-Blend plus.
3. For daily use, store at room temperature. Refrigerate bulk storage.
4. Redirect overhead light to prevent premature polymerization.
5. Resins can be sensitizing. Avoid repeated contact of uncured dental resinwith skin. Do not use on patients with a known sensitivity to acrylates or otherresins. If allergic reaction, dermatitis or rash develops, consult a physician.
6. Dispose of used tips and empty syringes properly.
7. Keep out of reach of children.
8. To avoid cross contamination, do not re-use tips.
9. Do not use after expiration date noted on containers.
10.To avoid cross contamination, disinfect and wipe syringe with an intermediatelevel disinfectant between uses and/or use a disposable syringe cover. If
syringe cover is used, remove tip, re-cap, and discard tip and syringe cover.
11.Keep Ultra-Blend plus out of heat/direct sunlight.

U

B nd P u

ES

Descripción:
Ultra-Blend™ plus es un liner (forro) y material de base de fotocurado, radiopaco, conadhesión a dentina, que contiene hidróxido de calcio e hidroxiapatita de calcio en una
base de dimetacrilato de uretano (UDMA). Ultra-Blend plus tiene una alta carga derelleno para minimizar la contracción y una mayor viscosidad para facilitar la sencillezde colocación. Puede ser modelado fácilmente con turbina o contraángulo (alta o baja
velocidad) y está disponible en tonos dentina y blanco opaco.
Indicaciones:
UltraBlend plus es auto-adherente y se utiliza como liner cavitario y/o como material debase previo a la colocación de materiales restauradores.
Instrucciones de uso:
A. General
1. Enrosque firmemente una punta Black Micro™ en la jeringa. Para preparaciones
mayores, utilice la punta Black Mini™.
2. Prepare la zona con un sistema adhesivo de calidad siguiendo las instrucciones
del fabricante.
3. Aplique una capa de 0,5-2mm de Ultra-Blend plus en la zona deseada.
NOTA: Cuando lo coloque en una zona de difícil acceso, dispense una pequeña
cantidad para formar una bolita en la misma punta. Utilice la punta como un
aplicador a bolita para evitar dispensar demasiado.
4. Fotopolimerice con la lámpara de fotopolimerización VALO™ LED en Potencia
Estándar durante 10 segundos (para lámparas con potencia <600mW/cm2
polimerice 20 segundos y con lámparas con una potencia >600mw/cm2
polimerice 10 segundos). NOTA: Para generar una base, comience con una
capa de 0,5mm y fotopolimerice. Siga con incrementos de 2mm, polimerizando
entre incrementos.
5. Ultra-Blend plus se adherirá a superficies preparadas con agentes adhesivos
dentinarios y a dentina seca.
B. Liner Dentinario
1. Enjuague la dentina expuesta con agua y seque con aire.
2. Aplique un sistema adhesivo de calidad siguiendo las instrucciones del fabricante.
3. Coloque Ultra-Blend plus en forma precisa y en la cantidad exacta que se desea
utilizando la punta Black Micro.
4. Fotopolimerice con la lámpara de fotopolimerización VALO™ LED en Potencia
Estándar durante 10 segundos (para lámparas con potencia <600mW/cm2
polimerice 20 segundos y con lámparas con una potencia >600mw/cm2
polimerice 10 segundos).
C. Base Protectora
1. Enjuague la dentina expuesta con agua y seque con aire.
2. Aplique un sistema adhesivo de calidad siguiendo las instrucciones del fabricante.
3. Si el área de la base es pequeña, enrosque una punta Black Micro en la jeringa;
utilice una punta Black Mini para zonas mayores.
4. Aplique Ultra-Blend plus y fotopolimerice con la lámpara de fotopolimerización
VALO™ LED en Potencia Estándar durante 10 segundos (para lámparas con
potencia <600mW/ cm2 polimerice 20 segundos y con lámparas con una potencia>600mw/cm2 polimerice 10 segundos).
5. Si la base final va a ser más gruesa que 2mm, aplique una capa inicial de 0,5mm(o menos) y fotopolimerice 10 segundos. Coloque capas adicionales de 2mm deespesor según sea necesario.
Nota: Ultra-Blend plus fotopolimeriza hasta una profundidad de 3mm.
D. Cobertor Pulpar
NOTA: Utilice Ultra-Blend plus en exposiciones cercanas a la pulpa (rosadas) y enpequeñas exposiciones mecánicas, no hiperémicas. En caso de exposiciones mayores
y/o pulpa hiperémica debería iniciarse una terapia endodóntica.
1. Aplique la solución antibacteriana Consepsis™ con la punta plástica Mini DentoInfusor™ durante 60 segundos utilizando un movimiento de fricción suave. No
enjuague. Seque suavemente con aire.
2. Aplique Ultra-Blend plus con la punta Black Micro sobre la dentina seca en elborde de la exposición y luego sobre la exposición o sobre la dentina rosada dela casi exposición.
NOTA: Minimice la cobertura de la dentina para maximizar la zona disponible para lastécnicas adhesivas dentinarias.
3. Fotopolimerice con la lámpara de fotopolimerización VALO™ LED en Potencia
Estándar durante 10 segundos (para lámparas con potencia <600mW/cm2
polimerice 20 segundos y con lámparas con una potencia >600mw/cm2
polimerice 10 segundos).
4. Aplique un sistema adhesivo de calidad siguiendo las instrucciones del fabricantey restaure con el producto de su elección.
E. Opacador
1. Asperice el metal con una fresa o mediante microabrasión.
2. Enjuague con agua el metal expuesto a ser opacado y séquelo con aire.
3. Aplique un sistema adhesivo de calidad siguiendo las instrucciones del fabricante
y fotopolimerice.
4. Aplique Ultra-Blend plus sobre el metal en incrementos, sin que ninguna capasea de un espesor superior a los 2mm. Fotopolimerice con la lámpara de
fotopolimerización VALO™ LED en Potencia Estándar durante 10 segundos (paralámparas con potencia <600mW/cm2 polimerice 20 segundos y con lámparas con
una potencia >600mw/cm2 polimerice 10 segundos).
5. Restaure con resina compuesta o el producto de su elección.
Precauciones:
ADVERTENCIA: El producto químico puede fraguar en la punta después de unas
horas. Para evitar atascos, retire la punta y vuelva a tapar con la tapa Luer Lock entre
usos. Verifique siempre el flujo de material antes de aplicar intraoralmente. Si nota
resistencia, reemplace la punta y vuelva a controlar. Utilice solamente las puntas
recomendadas.
1. Lea cuidadosamente y comprenda todas las instrucciones y la HDSM antes de utilizar
Ultra-Blend plus. Utilice sólo como se indica.
2. El clínico y el paciente deberían utilizar gafas de protección UV durante lapolimerización de Ultra-Blend plus.
3. Para uso diario, almacene a temperatura ambiente. Refrigere si almacena al
por mayor.
4. Modifique la posición de la luz operatoria para evitar una polimerización prematura.
5. Las resinas pueden sensibilizar. Evite el contacto repetido de la resina sin
fotopolimerizar con la piel. No utilice en pacientes con sensibilidad conocida aacrilatos y otras resinas. Si surge una reacción alérgica, dermatitis o un sarpullido,
consulte al médico.
6. Descarte adecuadamente las puntas usadas y las jeringas vacías
7. Mantenga fuera del alcance de los niños.
8. Para evitar la contaminación cruzada, no reutilice las puntas.
9. No utilice después de la fecha de vencimiento que figura en cada envase.
10. Para evitar la contaminación cruzada, desinfecte y limpie la jeringa con un
desinfectante de nivel intermedio entre usos y/o utilice las fundas descartables para
jeringas. Si se utiliza una funda para jeringa, desenrosque la punta, vuelva a tapar lajeringa y descarte la punta y la funda de la jeringa.
11. Mantenga Ultra-Blend plus lejos del calor/ luz solar directa.

m

U

B nd P uDE

Beschreibung:
Ultra-Blend™ plus ist ein lichthärtendes, röntgendichtes, an Dentin haftendes Liner- und
Unterfüllungsmaterial, das Kalziumhydroxid und Kalziumhydroxylapatit auf einer
Urethandimethacrylat-Basis (UDMA) enthält. Ultra-Blend ist reich an Füllstoffen, um
Schrumpfung zu minimieren und ist hochviskös für eine leichte Platzierung. Es kann leicht
mit einem Handstück mit hoher oder niedriger Geschwindigkeit in Form gebracht werden
und ist in den Farben Dentin und Opaque White verfügbar.
Anwendungsbereiche:
Ultra-Blend ist selbsthaftend und wird als Liner und/oder als Unterfüllungsmaterial vor
der Platzierung von restaurativem Material verwendet.
Gebrauchsanweisung:
A. Allgemeines
1. Befestigen Sie einen Black Micro™ Tip sicher auf der Spritze. Benutzen Sie bei
größeren Präparationen einen Black Mini™ Tip.
2. Bereiten Sie den Bereich mit einem Bonding-System guter Qualität gemäß den
Herstelleranweisungen vor.
3. Bringen Sie eine 0,5 - 2 mm dicke Schicht Ultra-Blend plus auf den gewünschten
Bereich auf.
HINWEIS: Wenn Sie in einem Bereich arbeiten, wo der Zugang schwierig ist, pressen
Sie eine kleine Menge heraus, die eine Kugel an dem Tip formt. Benutzen Sie den Tip
als Applikator für die Kugel, um zu verhindern, dass zu viel aufgebracht wird.
4. Lichthärten Sie mit der VALO™ LED Polymerisationsleuchte auf „Standard Power“
für 10 Sekunden (bei Leuchten mit einer Stärke von <600 mW/cm2 20 Sekundenaushärten, bei Leuchten mit einer Stärke von >600 mW/cm2 10 Sekunden).
HINWEIS: Beginnen Sie einen Aufbau mit einer Schicht von 0,5 mm und härten Sie
mit Licht. Fahren Sie mit weiteren Schichten von je 2 mm fort und härten Sie nachjeder Schicht.
5. Ultra-Blend plus haftet auf Oberflächen, die mit einem Dentinbonding behandelt
wurden, sowie auf trockenem Dentin.
B. Dentin-Liner
1. Spülen Sie das freiliegende Dentin mit Wasser und lufttrocknen Sie.
2. Bringen Sie auf den Bereich ein Bonding-System guter Qualität gemäß den
Herstelleranweisungen auf.
3. Applizieren Sie Ultra-Blend plus präzise und in der gewünschten Menge mit einem
Black Micro Tip.
4. Lichthärten Sie mit der VALO™ LED Polymerisationsleuchte auf „Standard Power“
für 10 Sekunden (bei Leuchten mit einer Stärke von <600 mW/cm2 20 Sekundenaushärten, bei Leuchten mit einer Stärke von >600 mW/cm2 10 Sekunden).
C. Schützende Unterfüllung
1. Spülen Sie das freiliegende Dentin mit Wasser und lufttrocknen Sie.
2. Bringen Sie auf den Bereich ein Bonding-System guter Qualität gemäß den
Herstelleranweisungen auf.
3. Wählen Sie für die Spritze einen Black Micro™ Tip, wenn der Bereich klein ist undeinen Black Mini™ Tip bei größeren Bereichen.
4. Applizieren Sie Ultra-Blend plus und lichthärten Sie mit der VALO™ LED
Polymerisationsleuchte auf „Standard Power“ für 10 Sekunden (bei Leuchten mit
einer Stärke von <600 mW/cm2 20 Sekunden aushärten, bei Leuchten mit einer
Stärke von >600 mW/cm2 10 Sekunden).
5. Wenn die endgültige Basis-Füllung stärker als 2 mm sein soll, tragen Sie als erste
Schicht 0,5 mm (oder weniger) auf und lichthärten Sie für 10 Sekunden. Fügen Sie
weitere Schichten von 2 mm Stärke hinzu, wie Sie sie benötigen.
Hinweis: Ultra-Blend plus kann bis zu einer Dicke von 3 mm lichtgehärtet werden.
D. Pulpenüberkappung
HINWEIS: Setzen Sie Ultra-Blend plus nahe der Pulpa (rosa) und bei kleiner,
mechanischer, nicht-hyperämischer Exposition ein. Bei größerer Exposition und/oder
hyperämischer Pulpa sollte eine endodontische Therapie eingeleitet werden.
1. Bringen Sie die antibakterielle Consepsis™ Lösung mittels eines Mini Dento-Infusor™
Tip aus Kunststoff mit sanften, bürstenden Bewegungen auf. Nicht spülen. Sanft
mit Luft trocknen.
2. Bringen Sie Ultra-Blend plus mit einem Black Micro™ Tip auf das trockene Dentin am
Rand der Exposition auf und ziehen Sie es dann über die Exposition oder über das
dünne (rosa) Dentin der nahezu exponierten Stelle.
HINWEIS: Minimieren Sie den Bereich des abgedeckten Dentins, um den Bereich zu
maximieren, der für die Dentin-Bonding-Techniken zur Verfügung steht.
3. Lichthärten Sie mit der VALO™ LED Polymerisationsleuchte auf „Standard Power“
für 10 Sekunden (bei Leuchten mit einer Stärke von <600 mW/cm2 20 Sekundenaushärten, bei Leuchten mit einer Stärke von >600 mW/cm2 10 Sekunden).
4. Bringen Sie auf den Bereich ein Bonding-System guter Qualität gemäß den
Herstelleranweisungen auf und restaurieren Sie mit einem Produkt Ihrer Wahl.
E. E. Als Opaker
1. Rauen Sie das Metall mit einem Schleifkörper oder durch Mikroabrasion auf.
2. Spülen Sie das freigelegte Metall, das mit Opaker bedeckt werden soll, mit Wasser
und lufttrocknen Sie.
3. Bringen Sie auf den Bereich ein Bonding-System guter Qualität gemäß den
Herstelleranweisungen auf und lichthärten Sie.
4. Bringen Sie Ultra-Blend plus in Schichten von höchstens 2 mm auf dem Metall auf.
Lichthärten Sie mit der VALO™ LED Polymerisationsleuchte auf „Standard Power“
für 10 Sekunden (bei Leuchten mit einer Stärke von <600 mW/cm2 20 Sekundenaushärten, bei Leuchten mit einer Stärke von >600 mW/cm2 10 Sekunden).
5. Restaurieren Sie mit Komposit oder einem Produkt Ihrer Wahl.
Vorsichtsmaßnahmen:
WARNUNG: Das Präparat kann in dem Tip nach ein paar Stunden aushärten. Um
das Verstopfen zu vermeiden, entfernen Sie den Tip und verschließen Sie zwischen
den Verwendungen mit Luer Lock Caps. Prüfen Sie stets vor jeder Benutzung den
Materialfluss, bevor Sie intraoral arbeiten. Wenn Sie Widerstand spüren, ersetzen Sie den
Tip und prüfen Sie erneut. Setzen Sie nur die empfohlenen Spritzenansätze ein.
1. Lesen und verstehen Sie vor der Benutzung von Ultra-Blend plus alle Anweisungenund Material-Sicherheitsdatenblätter sorgfältig.Nur nach Vorschrift benutzen.
2. Der Behandler und der Patient sollten UV-Augenschutz tragen, wenn Ultra-Blend
plus ausgehärtet wird.
3. Für täglichen Gebrauch bei Raumtemperatur lagern. Größere Mengen gekühlt lagern.
4. OP-Licht abblenden, um vorzeitige Polymerisation zu vermeiden.
5. 6.Kunststoffe können sensibilisieren. Vermeiden Sie wiederholten Kontakt von nicht
ausgehärtetem Dentalkunststoff mit der Haut. Nicht bei Patienten mit einer bekannten
Sensibilität auf Acrylate oder andere Kunststoffe anwenden. Falls eine allergische
Reaktion, Dermatitis oder ein Hautausschlag auftreten, kontaktieren Sie einen Arzt.
6. Entsorgen Sie benutzte Spritzenansätze und leere Spritzen fachgerecht.
7. Außerhalb der Reichweite von Kindern aufbewahren.
8. Um Kreuzkontamination zu vermeiden, benutzen Sie die Tips nicht mehrfach.
9. Nicht nach Ablauf des auf der Verpackung aufgedruckten Verfallsdatums benutzen.
10. Um Kreuzkontamination zu vermeiden, wischen Sie die Spritze ab und desinfizieren
Sie sie mit einem Flächendesinfektionsmittel zwischen den einzelnen Benutzungenund/oder verwenden Sie eine Einweg-Spritzenhülle. Wenn eine Spritzenhülle benutzt
wird, entfernen Sie den Tip, setzen Sie die Kappe wieder auf und entsorgen Sie Tip
und Spritzenhülle.
11. Ultra-Blend plus nicht Hitze oder direktem Sonnenlicht aussetzen.

mU

B nd P u

PT

Descrição:Ultra-Blend™ plus é um revestimento aderente à dentina e material base fotopolimerizável,radiopaco contendo hidróxido de cálcio e hidroxiapatita de cálcio numa base dedimetacrilato de uretano (UDMA). Ultra-Blend plus possui uma carga de preenchimento
elevada para reduzir a contracção e uma maior viscosidade para uma aplicação mais fácil.
O contorno é fácil com uma peça de mão de alta ou baixa velocidade e está disponível em
tonalidades de dentina e branco opaco.
Indicações:UltraBlend plus é auto-adesivo e é utilizado como revestimento de cavidade e/ou materialbase anteriormente à aplicação de materiais restauradores.
Instruções de utilização:A. Aspectos gerais
1. Aplicar a ponta Black Micro™ firmemente na seringa. Para preparações maiores,
utilizar uma ponta Black Mini™.
2. Preparar a área com um sistema adesivo de qualidade de acordo com as instruções
do fabricante.
3. Aplicar uma camada com 0,5-2 mm de espessura de Ultra-Blend plus na áreapretendida.
NOTA: Na aplicação numa área de acesso difícil, espremer uma pequena quantidadepara formar uma esfera na ponta. Utilizar a ponta como um aplicador esférico paraevitar a dispensa de uma quantidade excessiva.
4. Fotopolimerizar com luz de polimerização LED VALO™ com Potência Standard
durante 10 segundos (para luzes com uma saída <600mW/cm2 polimerizar durante
20 segundos ou para luzes com uma saída >600mw/cm2 polimerizar durante 10
segundos). NOTA: Para a formação da base, iniciar com uma camada de 0,5 mm deespessura e fotopolimerizar. Prosseguir com incrementos de 2 mm, polimerizando
entre cada incremento.
5. Ultra-Blend plus irá aderir às superfícies preparadas com adesivos dentinários eem dentina seca.
B. Revestimento dentinário
1. Enxaguar a dentina exposta com água e secar com ar.
2. Aplicar um sistema adesivo de qualidade de acordo com as instruções do fabricante.
3. Colocar o Ultra-Blend plus com precisão e com o volume exacto pretendido com
uma ponta Black Micro.
4. Fotopolimerizar com luz de polimerização LED VALO™ com Potência Standard
durante 10 segundos (para luzes com uma saída <600mW/cm2 polimerizar durante
20 segundos ou para luzes com uma saída >600mw/cm2 polimerizar durante
10 segundos).
C. Base protectora
1. Enxaguar a dentina exposta com água e secar com ar.
2. Aplicar um sistema adesivo de qualidade de acordo com as instruções do fabricante.
3. Aplicar uma ponta Black Micro na seringa se a área da base for reduzida; utilizar
uma ponta Black Mini para áreas maiores.
4. Aplicar Ultra-Blend plus e fotopolimerizar com luz de polimerização LED VALO™
com Potência Standard durante 10 segundos (para luzes com uma saída <600mW/
cm2 polimerizar durante 20 segundos ou para luzes com uma saída >600mw/cm2
polimerizar durante 10 segundos).
5. Para que a base final tenha uma espessura superior a 2 mm, aplicar uma primeira
camada com 0,5 mm (ou menos) de espessura e fotopolimerizar durante 10
segundos. Adicionar camadas adicionais com 2 mm de espessura, conforme
necessário.
Nota: Ultra-Blend plus fotopolimeriza até uma profundidade de 3 mm.
D. Capeamento pulpar
NOTA: Utilizar Ultra-Blend plus para polpa próxima (cor-de-rosa) e para exposição
mecânica pequena não -hiperémica. Para uma exposição maior e/ou polpa hiperémica,
deverá ser iniciada terapia endodôntica.
1. Aplicar a solução antibacteriana Consepsis™ com ponta plástica Mini Dento-Infusor™
durante 60 segundos utilizando uma ligeira acção de fricção. Não enxaguar. Secar
cuidadosamente com ar.
2. Aplicar Ultra-Blend plus com a ponta Black Micro na dentina seca no rebordoda exposição e depois sobre a exposição ou sobre a dentina cor-de-rosa da
exposição próxima.
NOTA: Minimizar a cobertura da dentina para maximizar a área disponível para as
técnicas de adesão dentinária.
3. Fotopolimerizar com luz de polimerização LED VALO™ com Potência Standard
durante 10 segundos (para luzes com uma saída <600mW/cm2 polimerizar durante
20 segundos ou para luzes com uma saída >600mw/cm2 polimerizar durante
10 segundos).
4. Aplicar um sistema adesivo de qualidade de acordo com as instruções do fabricantee restaurar com o produto da sua escolha.
E. Agente opacidante
1. Desbastar o metal com broca ou através de microabrasão.
2. Enxaguar o metal exposto a opacidar com água e secar com ar.
3. Aplicar um sistema adesivo de qualidade de acordo com as instruções do fabricantee fotopolimerizar.
4. Aplicar Ultra-Blend plus sobre o metal em incrementos com camadas com
espessuras não superiores a 2 mm. Fotopolimerizar com luz de polimerização LED
VALO™ com Potência Standard durante 10 segundos (para luzes com uma saída<600mW/cm2 polimerizar durante 20 segundos ou para luzes com uma saída>600mw/cm2 polimerizar durante 10 segundos).
5. Restaurar com compósito ou produto à sua escolha.
Precauções:ADVERTÊNCIA: O químico pode polimerizar na ponta após algumas horas. Para evitar aobstrução, retirar a ponta e fechar novamente as tampas Luer Lock entre cada utilização.
Verificar sempre o fluxo do material antes da aplicação intra-oral. Em caso de resistência,
substituir a ponta e verificar novamente. Utilizar apenas as pontas recomendadas.
1. Ler atentamente e compreender todas as instruções e fichas de segurança antes dautilização do Ultra-Blend plus. Utilizar apenas conforme indicado.
2. O médico e paciente devem usar proteção ocular UV durante a polimerização do
Ultra-Blend plus.
3. Para a utilização diária guardar à temperatura ambiente. Refrigerar paraarmazenamento a granel.
4. Redireccionar o foco odontológico para evitar a polimerização prematura.
5. As resinas podem ser irritantes. Evitar o contacto repetido de resina dentária não
polimerizada com a pele. Não utilizar em pacientes com sensibilidade conhecida aacrilatos ou outras resinas. Em caso de reação alérgica, dermatite ou se surgir erupção
cutânea, consultar um médico.
6. Descartar as pontas usadas e as seringas vazias correctamente.
7. Manter fora do alcance das crianças.
8. Para evitar a contaminação cruzada, não reutilizar as pontas.
9. Não utilizar após a data de validade anotada no recipiente.
10. Para evitar a contaminação cruzada, desinfectar e limpar a seringa com um
desinfectante de nível intermédio entre utilizações e/ou utilizar uma cobertura deseringa descartável. Se for utilizada a cobertura da seringa, remover a ponta, recolocar
a tampa e descartar a ponta e a cobertura da seringa.
11. Manter o Ultra-Blend plus protgido contra calor/luz solar directa.

U

B nd P um

FR

Description:
Ultra-Blend™ plus est un produit isolant adhérant à la dentine, photopolymérisable,
opaque aux radiographies et utilisé comme matériau de base contenant de l’hydroxide
de calcium et de hydroxyapatite de calcium dans une base au diméthacrylate d’uréthane.
Ultra-Blend plus a une charge élevée d’agent inerte pour minimer le rétrécissement eta une haute viscosité pour en faciliter le placement. Il peut être facilement façonné avec
une pièce à main à grande ou basse vitesse et est disponible en teintes pour la dentine
ou blanc opaque.
Indications:
Ultra-Blend plus est auto-adhésif et est utilisé comme isolant de caries et/ou commematériau de base avant le placement des matériaux de restauration.
Mode d’emploi:
A. Général
1. Fixez fermement l’embout Black Micro™ sur la seringue. Pour des préparations plus
importantes, utilisez l’embout Black Mini™.
2. Préparez la zone avec un système de liaison de qualité en suivant les instructions
du fabricant.
3. Appliquez une couche de 0,5 à 2 mm d’Ultra-Blend plus sur la zone choisie.
NOTE : Lors de l’application du produit sur une zone d’accès difficile, exprimez une
petite quantité pour former une boule au bout de l’embout. Utilisez l’embout comme unapplicateur boule pour éviter une surapplication.
4. Photopolymérisez avec la lampe de polymérisation VALO™ LED en mode de puissancestandard pendant 10 secondes (pour les lampes ayant une puissance de sortie<600mW/ cm2 polymérisez pendant 20 secondes ou pour les lampes ayant une
puissance de sortie >600mw/cm2 polymérisez pendant 10 secondes).
NOTE : Pour établir une base, démarrez avec une couche de 0,5 mm et
photopolymérisez. Continuez avec des couches suivantes de 2 mm en polymérisantentre chaque couche.
5. Ultra-Blend plus se liera aux surfaces préparées avec les agents de liaison pourdentine et sur la dentine sèche.
B. Isolant pour dentine
1. Rincez la dentine exposée à l’eau et sèchez à l’air.
2. Appliquez un système de liaison de qualité en suivant les instructions du fabricant.
3. Placez l’Ultra-Blend plus précisément et dans le volume exact souhaité à l’aide del’embout Black Micro.
4. Photopolymérisez avec la lampe de polymérisation VALO™ LED en mode de puissancestandard pendant 10 secondes (pour les lampes ayant une puissance de sortie<600mW/ cm2 polymérisez pendant 20 secondes ou pour les lampes ayant une
puissance de sortie >600mw/cm2 polymérisez pendant 10 secondes).
C. Base protectrice
1. Rincez la dentine exposée à l’eau et sèchez à l’air.
2. Appliquez un système de liaison de qualité en suivant les instructions du fabricant.
3. Fixez l’embout Black Micro sur la seringue si la zone de la base est restreinte ; utilisez
un embout Black Mini pour les zones plus larges.
4. Appliquez l’Ultra-Blend plus et photopolymérisez avec la lampe de polymérisation
VALO™ LED en mode de puissance standard pendant 10 secondes (pour les lampes
ayant une puissance de sortie <600mW/cm2 polymérisez pendant 20 secondes ou
pour les lampes ayant une puissance de sortie >600mw/cm2 polymérisez pendant
10 secondes).
5. Si la base finale doit faire plus de 2mm d’épaisseur, appliquez 0,5 mm (ou moins) en
première couche et photopolymérisez pendant 10 secondes. Ajoutez des couches
supplémentaires de 2 mm d’épaisseur si nécessaire.
Note : Ultra-Blend plus se photopolymérise à une profondeur de 3 mm.
D. Coiffage pulpaire
NOTE : Utilisez l’Ultra-Blend plus pour les quasi-expositions de la pulpe (rose) et lespetites expositions mécaniques non-hyperémiques.En cas d’expositions plus grandes et/ou en présence de pulpe hyperémique, un traitement endodontique devrait être initié.
1. Appliquez la solution antibactérienne Consepsis™ à l’aide de l’embout en plastique
Mini Dento-Infusor™ pendant 60 secondes en utilisant un doux mouvement defrottement. Ne rincez pas. Sèchez à l’air doux.
2. Appliquez l’Ultra-Blend plus à l’aide de l’embout Black Micro sur la dentine sèche
en bordure de l’exposition puis sur l’exposition ou sur la fine dentine rose de laquasi-exposition.
NOTE : Minimisez la couverture de la dentine pour maximisez la zone disponible aux
techniques de liaison sur la dentine.
3. Photopolymérisez avec la lampe de polymérisation VALO™ LED en mode de puissancestandard pendant 10 secondes (pour les lampes ayant une puissance de sortie<600mW/ cm2 polymérisez pendant 20 secondes ou pour les lampes ayant une
puissance de sortie >600mw/cm2 polymérisez pendant 10 secondes).
4. Appliquez un système de liaison de qualité en suivant les instructions du fabricant et
restaurez avec le produit de votre choix.
E. Opaqueur
1. Grattez le métal à l’aide d’une fraise ou par microabrasion.
2. Rincez à l’eau le métal exposé à rendre opaque et sèchez à l’air.
3. Appliquez un système de liaison de qualité en suivant les instructions du fabricant
et photopolymérisez.
4. Appliquez l’Ultra-Blend plus sur le métal par incrément et dont aucune couche
ne dépasse 2mm d’épaisseur. Photopolymérisez avec la lampe de polymérisation
VALO™ LED en mode de puissance standard pendant 10 secondes (pour les lampes
ayant une puissance de sortie <600mW/cm2 polymérisez pendant 20 secondes ou
pour les lampes ayant une puissance de sortie >600mw/cm2 polymérisez pendant
10 secondes).
5. Restaurez avec le composite ou le produit de votre choix.
Precauzioni:
AVERTISSEMENT : Le produit chimique peut prendre dans l’embout au bout de quelques
heures. Pour éviter un bouchon, enlevez l’embout et rebouchez-le avec les capuchons
Luer Lock entre chaque utilisation. Vérifiez toujours l’écoulement du matériau avant del’appliquer en bouche. En cas de résistance, remplacez l’embout et revérifiez. N’utilisez queles embouts recommandés.
1. Lisez et comprenez toutes les instructions et les fiches techniques santé-sécurité avantd’utiliser le produit. A utiliser uniquement selon les indications.
2. Le clinicien et le patient devraient porter des lunettes de protection contre les UV
pendant la polymérisation de l’Ultra-Blend plus.
3. Pour une utilisation quotidienne, entreposez à température ambiante. Réfrigérez lestock en vrac.
4. Redirigez le plafonnier pour éviter une polymérisation prématurée.
5. Les résines peuvent provoquer une sensibilisation. Evitez un contact répété entre lapeau et la résine dentaire non polymérisée. Ne pas utiliser sur des patients ayant
une sensibilité connue aux acrylates et autres résines. Si une réaction allergique, unedermatite ou une éruption cutanée se développe, consultez un médecin.
6. Jetez les embouts usagés et les seringues vides de façon appropriée.
7. Tenir hors de portée des enfants.
8. Pour éviter la contamination croisée, ne réutilisez pas les embouts.
9. Ne pas utiliser après la date d’expiration notée sur les récipients.
10. Pour éviter la contamination croisée, désinfectez et essuyez les seringues avec undésinfectant de niveau intermédiaire entre chaque utilisation et/ou utilisez un protègeseringue jetable. Si un protège-seringue est utilisé, enlevez embout, rebouchez et jetez
l’embout et le protège-seringue.
11. Gardez l’Ultra-Blend plus à l’écart de la chaleur/de la lumière solaire directe.

mU

B nd P um

SV

Beskrivning:
Ultra-Blend™ plus är en ljushärdad, strålsäker, dentinfästande skyddsfilm ochbasmaterial som innehåller kalciumhydroxid och kalciumhydroxyapatit i en
uretandimetacrylatbas (UDMA). Ultra-Blend plus har en högre fyllnadsbelastning
för att minimera krympning och en högre viskositet för att underlätta placeringen.
Den kan med lätthet formas med ett handstycke med hög eller låg hastighet ochfinns i dentin och opak vit nyans.
Indikationer:
Ultra-Blend plus är självhäftande och används som kavitetsskyddsfilm och/ellerbasmaterial före placering av tandfyllnadsmaterial.
Bruksanvisning:
A. Allmänt
1. Fäst Black Micro™-spetsen stadigt på sprutan. Använd Black Mini™-spets för
större preparationer.
2. Förbered området med ett kvalitetsbondingssystem enligt tillverkarens
anvisningar.
3. Applicera ett 0,5-2 mm lager Ultra-Blend plus på det önskade området.
OBS: Tryck ut en liten mängd så att en boll formas på spetsen vid applicering
på ett svårtillgängligt område. Använd spetsen som en bollapplikator för att
undvika överdispensering.
4. Ljushärda med VALO™ LED härdningsljus i Standard effekt-läge i 10 sekunder(20 sekunder för ljus med en energiutveckling < 600mW/cm2 eller 10
sekunder för ljus med en energiutveckling > 600mW/cm2).
OBS: För basuppbyggnad, börja med ett 0,5 mm lager och ljushärda. Följ meden 2 mm stegvis ökning, härda mellan varje ökning.
5. Med Ultra-Blend plus sker bonding till ytor som preparerats med
dentinbondingsmedel och till torr dentin.
B. Dentin skyddsfilm
1. Skölj utsatt dentin med vatten och lufttorka.
2. Applicera ett kvalitetsbondingssystem enligt tillverkarens anvisningar.
3. Placera en exakt mängd Ultra-Blend plus noggrant med en Black Micro-spets.
4. Ljushärda med VALO™ LED härdningsljus i Standard effekt-läge i 10 sekunder(20 sekunder för ljus med en energiutveckling < 600mW/cm2 eller 10
sekunder för ljus med en energiutveckling > 600mW/cm2).
C. Skyddande bas
1. Skölj utsatt dentin med vatten och lufttorka.
2. Applicera ett kvalitetsbondingssystem enligt tillverkarens anvisningar.
3. Fäst en Black Micro-spets på sprutan om basområdet är litet; använd en Black
Mini-spets för större områden.
4. Applicera Ultra-Blend plus och ljushärda med VALO™ LED härdningsljus
i Standard effekt-läge i 10 sekunder (20 sekunder för ljus med enenergiutveckling < 600mW/cm2 eller 10 sekunder för ljus med enenergiutveckling > 600mW/cm2).
5. Om den slutliga basen ska vara tjockare än 2 mm, applicera 0,5 mm (eller
mindre) som första lager och ljushärda i 10 sekunder. Lägg till ytterligare
lager av en tjocklek på 2 mm enligt behov.
Obs: Ultra-Blend plus ljushärdas till ett djup på 3 mm.
D. Pulpskydd
OBS: Använd Ultra-Blend plus nära pulpa (ljusröd) och för liten mekanisk
hyperemfri exponering. Endodontisk behandling ska påbörjas om större
exponering och/eller om hyperemisk pulpa.
1. Applicera Consepsis™ antibakteriell lösning med en Mini Dento-Infusor™-spets
i 60 sekunder med en milt skrubbande rörelse. Skölj inte. Lufttorka försiktigt.
2. Applicera Ultra-Blend plus med en Black Micro-spets på det torra dentinet vid
kanten av exponeringen sedan över exponeringen eller det tunna ljusröda
dentinet nära exponeringen.
OBS: Minimera täckandet av dentin så att området som är tillgängligt för
dentinbondingtekniker maximeras.
3. Ljushärda med VALO™ LED härdningsljus i Standard effekt-läge i 10 sekunder(20 sekunder för ljus med en energiutveckling < 600mW/cm2 eller 10
sekunder för ljus med en energiutveckling > 600mW/cm2).
4. Applicera ett kvalitetsbondingssystem enligt tillverkarens anvisningar och
restorera med den valda produkten.
E. Genomskinlighet
1. Gör metall grövre med borr eller mikroabrasion.
2. Skölj utsatt metall som ska bli opakt med vatten och lufttorka.
3. Applicera ett kvalitetsbondingssystem enligt tillverkarens anvisningar och
ljushärda.
4. Applicera Ultra-Blend plus på metall stegvis med lager som inte är tjockareän 2 mm. Ljushärda med VALO™ LED härdningsljus i Standard effekt-läge i 10
sekunder (20 sekunder för ljus med en energiutveckling < 600mW/cm2 eller
10 sekunder för ljus med en energiutveckling > 600mW/cm2).
5. Restorera med den valda kompositen eller produkten.
Försiktighetsåtgärder:
VARNING: Det kemiska ämnet kan härdas i spetsen efter några timmar. Avlägsna
spetsen och lägg åter lock på med Luer Lock-lock mellan användningarna för
att undvika att spetsen täpps till. Kontrollera alltid materialflödet före intraoral
applicering. Om det finns motstånd ska spetsen bytas ut och flödet kontrolleras på
nytt. Använd endast rekommenderade spetsar.
1. Läs noga igenom och förstå alla instruktioner och materialsäkerhetsbladet före
användning av Ultra-Blend plus. Använd endast i enlighet med anvisningarna.
2. Läkaren och patienten ska använda UV-skyddsglasögon vid härdning av
Ultra-Blend plus.
3. Förvaras i rumstemperatur för daglig användning. Förvara stora partier ikylskåp.
4. Rikta bort ljuset ovanför att förhindra för tidig polymerisering.
5. Harts kan vara sensibiliserande. Undvik att huden kommer i upprepad kontakt
med ohärdat dentalt harts. Använd inte på patienter med känd känslighet
för akrylater eller andra hartser. Om allergisk reaktion, dermatit eller utslag
utvecklas bör läkare vidtalas.
6. Kasta bort använda spetsar och tomma sprutor på korrekt sätt.
7. Förvaras utom räckhåll för barn.
8. För att undvika korskontamination ska spetsarna inte återanvändas.
9. Använd inte efter utgångsdatum som finns angivet på behållarna.
10. Desinficera och torka av sprutan och blandningsspetsen med ett intermediärt
desinficeringsmedel mellan användningarna och/eller använd ettengångssprutskydd för att undvika korskontamination. Avlägsna spetsen, sätt
tillbaka locket och kasta spetsen och sprutskyddet om sprutskydd används.
11. Utsätt inte Ultra-Blend plus för värme/direkt solljus.

mU

B nd P um

NL

Beschrijving:
Ultra-Blend™ plus is een lichtuithardend, radiopake, zich aan het dentine hechtendeliner en basismateriaal dat calciumhydroxide en calciumhydroxylapatiet bevat in een
urethaandimethacrylaat (UDMA)-basis. Ultra-Blend plus heeft een hoog vulpercentage om
krimping tot een minimum te beperken en een hogere viscositeit om het aanbrengen tevergemakkelijken. De contour kan eenvoudig worden aangepast met een handstuk met
hoog of laag toerental en het product is beschikbaar in dentine en opake witte tinten.
Indicaties:
Ultra-Blend plus is zelfhechtend en wordt gebruikt als een caviteitsliner en/ofbasismateriaal vóór het plaatsen van restauratiematerialen.
Gebruiksaanwijzing:
A. Algemeen
1. Maak de Black Micro™ tip goed vast op de spuit. Gebruik voor grotere preparatiesde Black Mini™ tip.
2. Prepareer de plaats met een hoogwaardig bondingsysteem volgens de instructies
van de fabrikant.
3. Breng een 0,5-2mm laag Ultra-Blend plus aan op de gewenste plaats.
NB: Bij moeilijk toegankelijke plaatsen kunt u een kleine hoeveelheid eruit drukken
zodat zich bij de tip een bol vormt. Gebruik de tip als een bolapplicator om tevoorkomen dat u teveel uit de spuit drukt.
4. Hard uit met een VALO™ LED Curing Light bij Standaardvermogen gedurende 10
seconden (hard bij lampen met een vermogen < 600 mW/cm2 20 seconden uit of bij
lampen met een vermogen > 600 mW/cm2 gedurende 10 seconden).
NB: Om een basis op te bouwen, moet u starten met een 0,5 mm laag en deze danuitharden. Breng daarna steeds een laag van 2 mm aan en hard deze telkens na het
aanbrengen uit.
5. Ultra-Blend plus hecht aan oppervlakken die met dentinehechtmiddelen zijn
geprepareerd en aan droog dentine.
B. Dentineliner
1. Spoel het blootliggende dentine af met water en droog het met lucht.
2. Breng een hoogwaardig bondingsysteem aan volgens de instructies van defabrikant.
3. Breng Ultra-Blend plus nauwkeurig en in de gewenste hoeveelheid aan met een
Black Micro tip.
4. Hard uit met een VALO™ LED Curing Light bij Standaardvermogen gedurende 10
seconden (hard bij lampen met een vermogen < 600 mW/cm2 20 seconden uit of bij
lampen met een vermogen > 600 mW/cm2 gedurende 10 seconden).
C. Beschermende basis
1. Spoel het blootliggende dentine af met water en droog het met lucht.
2. Breng een hoogwaardig bondingsysteem aan volgens de instructies van defabrikant.
3. Bevestig een Black Micro tip aan de spuit als de basisplaats klein is; gebruik een
Black Mini tip voor grotere plaatsen.
4. Hard Ultra-Blend plus uit met een VALO™ LED Curing Light bij Standaardvermogen
gedurende 10 seconden (hard bij lampen met een vermogen < 600 mW/cm2
20 seconden uit of bij lampen met een vermogen > 600 mW/cm2 gedurende
10 seconden).
5. Als de definitieve basis dikker moet zijn dan 2 mm, moet u eerst een 0,5 mm (ofdunnere) laag aanbrengen en deze gedurende 10 seconden uitharden. Brengdaarna extra lagen van 2 mm dikte aan.
NB: Ultra-Blend plus kan tot 3 mm diepte worden uitgehard.
D. Pulpaoverkapping
NB: Gebruik Ultra-Blend plus voor plaatsen in de buurt van de pulpa (roze) en
kleine mechanische niet-hyperemische blootliggende gebitselementen. Bij grotere
blootliggende gebitselementen en/of bij hyperemische pulpa is een endodontischebehandeling noodzakelijk.
1. Breng Consepsis™ antibacteriële oplossing aan met de kunststof Mini Dento-Infusor™
tip gedurende 60 seconden met een licht schrobbende beweging. Niet afspoelen.
Voorzichtig met lucht drogen.
2. Breng Ultra-Blend plus met de Black Micro tip aan op het droge dentine aan de rand
van het blootliggende gebitselement en dan over het blootliggende gebitselementen over het dunne roze dentine van het nabijgelegen blootliggende gebitselement.
NB: Bedek het dentine zo weinig mogelijk zodat u de dentinebondingtechnieken
maximaal kunt gebruiken.
3. Hard uit met een VALO™ LED Curing Light bij Standaardvermogen gedurende 10
seconden (hard bij lampen met een vermogen < 600 mW/cm2 20 seconden uit of bij
lampen met een vermogen > 600 mW/cm2 gedurende 10 seconden).
4. Breng een hoogwaardig bondingsysteem volgens de instructies van de fabrikant en
breng de gewenste restauratie aan.
E. Opaquer
1. Ruw het metaal op met een frees of microabrasie.
2. Spoel het blootliggende metaal waarop de opaquer moet worden aangebracht af
met water en droog het met lucht.
3. Breng een hoogwaardig bondingsysteem aan volgens de instructies van defabrikant.
4. Breng Ultra-Blend plus aan op het metaal in achtereenvolgende lagen, waarbij elkelaag niet dikker mag zijn dan 2 mm. Hard uit met een VALO™ LED Curing Light bij
Standaardvermogen gedurende 10 seconden (hard bij lampen met een vermogen< 600 mW/cm2 20 seconden uit of bij lampen met een vermogen > 600 mW/cm2
gedurende 10 seconden).
5. Restaureer met het door u gekozen composiet of product.
Voorzorgsmaatregelen:
WAARSCHUWING: chemische stof kan na een paar uur in de tip uitharden. Om verstopping
te voorkomen, verwijdert u na elk gebruik de tip en brengt u de Luer Lock-dop weer
aan. Controleer altijd de stroming van het materiaal voordat u het intraoraal gebruikt.
Als u weerstand voelt, vervangt u de tip en controleert u opnieuw. Gebruik uitsluitend
aanbevolen tips.
1. Lees en begrijp het veiligheidsinformatieblad (SDS) en alle instructies zorgvuldig
voordat u het product gebruikt. Uitsluitend volgens de gebruiksaanwijzing gebruiken.
2. Tandarts en patiënt moeten UV-oogbescherming dragen tijdens het uitharden van
Ultra-Blend plus.
3. Bewaar het product bij dagelijks gebruik bij omgevingstemperatuur. Vries grote
hoeveelheden in.
4. Verdraai de operatielamp om vroegtijdige polymerisatie te voorkomen.
5. Harsen kunnen tot overgevoeligheid leiden. Voorkom herhaaldelijk contact tussen deniet-uitgeharde dentale hars en de huid. Niet gebruiken bij patiënten van wie bekend
is dat zij gevoelig zijn voor acrylaten of andere harsen. Raadpleeg een arts bij een
allergische reactie, dermatitis of huiduitslag.
6. Gooi gebruikte tips en lege spuiten op de juiste manier weg.
7. Buiten bereik van kinderen houden.
8. Om kruisbesmetting te voorkomen, mogen de tips niet opnieuw worden gebruikt.
9. Niet gebruiken na de uiterste gebruiksdatum op de verpakking.
10. Om kruisbesmetting te voorkomen, moet de spuit tussen patiënten met een
middelmatig desinfectiemiddel worden gedesinfecteerd en afgeveegd en/of moetu een voor eenmalig gebruik bestemde spuitdop gebruiken. Als een spuitdop wordt
gebruikt, moet u de tip verwijderen, de dop weer aanbrengen en de spuitdop
weggooien.
11. Stel Ultra-Blend plus niet bloot aan hitte/direct zonlicht.

UæB nd P u

m

U

M

IT

M

W

W

M
W

W

M

M

W

W

M
M

W

W

W

W

DA

Beskrivelse:
Ultra-Blend™ plus er et lyspolymeriserbart, radiopakt bunddæknings- og
linermateriale, der adhæreret til dentin og indeholder calciumhydroxid og
calciumhydroxyapatit i en urethandimethacrylat (UDMA)-base. Ultra-Blend plus har
et højt filler-indhold for at minimere krybning og en højere viskositet for at letteplacering. Det kan nemt kontureres med et høj- eller lav-hastigheds håndstykke og fåsi dentin og opake hvide farver.
Indikationer:
Ultra-Blend plus er selvadhærerende og anvendes som en kavitetsliner og/eller
bunddækningsmateriale inden placering af fyldningsmaterialer.
Brugsanvisning:
A. Generelt
1. Sæt en Black Micro™ spids fast på sprøjten. Til større præparationer anvendes
Black Mini™ spids.
2. Præparer området med et adhæsivsystem af god kvalitet ifølge producentens
vejledning.
3. Læg et 0,5-2 mm lag Ultra-Blend plus på det ønskede område.
BEMÆRK: Ved anvendelse i et område, der er svært tilgængeligt, kan der trykkes
en smule materiale ud, så der dannes en kugle på spidsen. Brug spidsen som en
kuglestopper for at forhindre overdosering.
4. Lyspolymeriser med VALO™ LED polymeriseringslampe på Standard Power i 10
sekunder (for lamper med udgang < 600 mW/cm2 polymeriseres i 20 sekunder,eller for lamper med udgang > 600 mw/cm2 polymeriseres i 10 sekunder).
BEMÆRK: Ved opbygning af bunddækningsmateriale skal der startes med et lagpå 0,5 mm lag, der lyspolymeriseres. Hvert efterfølgende lag skal være 2 mm tyktog hvert lyspolymeriseres.
5. Ultra-Blend plus adhærerer til overflader, der er præpareret med etdentinadhæsiv.
B. Liner
1. Skyl eksponeret dentin med vand og tør efter med luft.
2. Anvend et adhæsivsystem af god kvalitet ifølge producentens vejledning.
3. Applicer Ultra-Blend plus præcist og i den nøjagtigt ønskede mængde med
en Black Micro-spids.
4. Lyspolymeriser med VALO™ LED polymeriseringslampe på Standard Power i 10
sekunder (for lamper med udgang < 600 mW/cm2 polymeriseres i 20 sekunder,eller for lamper med udgang > 600 mw/cm2 polymeriseres i 10 sekunder).
C. Bunddækningsmateriale
1. Skyl eksponeret dentin med vand og lad det lufttørre.
2. Anvend et adhæsivsystem af god kvalitet ifølge producentens vejledning.
3. Sæt en Black Micro-spids fast på sprøjten, hvis området er småt, og brug enBlack Mini-spids til større områder.
4. Applicer Ultra-Blend og lyspolymeriser med VALO™ LED polymeriseringslampepå Standard Power i 10 sekunder (for lamper med udgang < 600 mW/cm2
polymeriseres i 20 sekunder, eller for lamper med udgang > 600 mw/cm2
polymeriseres i 10 sekunder).
5. Hvis det endelige bunddækningsmateriale skal være tykkere end 2 mm, skal derlægges 0,5 mm (eller mindre) som det første lag, og det skal lyspolymeriseres i
10 sekunder. Der kan lægges flere lag a 2 mm efter behov.
Bemærk: Ultra-Blend plus lyspolymeriseres til en dybde af 3 mm.
D. PulpaoverkapningBEMÆRK: Brug Ultra-Blend plus nær pulpa (lyserød) og til små, mekaniske ikkehyperæmiske blottelser. Hvis der forekommer større blottelse, og/eller hvis pulpa
er hyperæmisk, skal der udføres endodontisk behandling.
1. Applicer Consepsis™ antibakterielt middel med en Mini Dento-Infusor™ spids
af plastik i 60 sekunder med en nænsom, skrubbende bevægelse. Der må ikke
skylles. Tør forsigtigt med luft.
2. Applicér Ultra-Blend plus med Black Micro-spids på den tørre dentin på kanten
af pulpablottelsen og derefter over blottelsen eller over den tynde, pulpanære,lyserøde dentin.
BEMÆRK: Dæk så lidt af dentinen som muligt for at maksimere det dentinområde,der kan anvendes til adhæsion.
3. Lyspolymeriser med VALO™ LED polymeriseringslampe på Standard Power i 10
sekunder (for lamper med udgang < 600 mW/cm2 polymeriseres i 20 sekunder,eller for lamper med udgang > 600 mw/cm2 polymeriseres i 10 sekunder).
4. Anvend et adhæsivsystem af god kvalitet ifølge producentens vejledning, og
restaurer med et egnet materiale.
E. Opakmasse
1. Gør metallet ru med et bor eller mikroabrasion.
2. Skyl det blottede metal med vand, og tør med luft.
3. Anvend et adhæsivsystem af god kvalitet ifølge producentens vejledning, og
lyspolymeriser det.
4. Applicér Ultra-Blend plus lagvist over metallet. Lagene må ikke være tykkere
end 2 mm. Lyspolymeriser med VALO™ LED polymeriseringslampe på Standard
Power i 10 sekunder (for lamper med udgang < 600 mW/cm2 polymeriseres
i 20 sekunder, eller for lamper med udgang > 600 mw/cm2 polymeriseres i
10 sekunder).
5. Brug komposit- eller andet materiale til reparationen.
Forholdsregler:
ADVARSEL: Materialet kan afbinde i spidsen efter et par timer. For at undgå tilstopningskal spidsen tages af, og en luerlock-hætte sættes på mellem hver anvendelse.
Kontroller altid flowet inden intraoral anvendelse. Hvis der mødes modstand, skal
spidsen udskiftes og kontrolleres igen. Der må kun anvendes anbefalede spidser.
1. Alle vejledninger og sikkerhedsdatablade skal læses nøje igennem inden
anvendelse af Ultra-Blend plus. Må kun anvendes som angivet.
2. Tandlæger, klinikassistenter og patienter skal bære UV-beskyttelsesbriller ved
polymerisering af Ultra-Blend plus.
3. Opbevares ved stuetemperatur til daglig anvendelse, men lagerbeholdningen skal
opbevares i køleskab.
4. Vend lamperne i en anden retning for at forhindre præmatur polymerisering.
5. Resinmaterialer kan være sensibiliserende. Undgå gentagen kontakt mellem
uafbundne dentale resinmaterialer og hud. Må ikke anvendelse på patienter med
kendt følsomhed overfor acrylater eller andre resinmaterialer. Hvis der opstår enallergisk reaktion, dermatitis eller eksem, skal der søges læge.
6. Bortskaf de brugte spidser og tomme sprøjter på korrekt vis.
7. Opbevares utilgængeligt for børn.
8. Spidserne må ikke genbruges for at undgå krydskontaminering.
9. Må ikke anvendes efter udløbsdatoen på beholderne.
10. Desinficér og tør sprøjten af med et mellemstærkt desinfektionsmiddel mellem
hver brug for at undgå krydskontaminering, eller brug en hygiejnepose over
sprøjten. Hvis der anvendes et sprøjtehylster, fjernes spidsen, hætten sættes påigen, og spidsene og sprøjtehylstret bortskaffes.
11. Ultra-Blend plus skal holdes væk fra varme/direkte sollys.

B nd P u

Descrizione:
Ultra-Blend™ plus è un liner/fondino di cavità con proprietà adesive verso la dentina,
fotopolimerizzabile e radiopaco contenente idrossido di calcio e idrossiapatite dicalcio in una base di dimetacrilato di uretano (UDMA). È caratterizzato da un alto
carico di riempitivo per ridurre la contrazione e da un’elevata viscosità che ne facilita
l’applicazione. Può essere facilmente sagomato mediante un manipolo ad alta o bUm

B nd P u

FI

M

M

M
M

W

W

M

M
W

M

M
W

W

W

M
M
M
M
W

W

W

W

m

m
mm
m

m
m

m

m

m

m
m

mm

m

m

m

m

m

m
m

m

m
m

m

m
m

m
m

Ultra-Blend™ Plus

AR LOT

REF

EN - Lot Number
DE - Chargennummer
FR - Numéro de lot
NL - Lotnummer
IT - Numero di lotto
ES - Número de lotePT - Número do loteSV - Partinummer
DA - Varenummer
FI - Eränumero
EL - Αριθμός παρτίδας
NO - Varenummer
PL - Numer partii
RU - Номер партии
TR - Lot Numarası
AR - ‫رقم الشحنة‬TH - หมายเลขล็็อต
KO - 품목 번호
ZH - 批号

EN - For professional use only
DE - Nur zur Anwendung durch den Zahnarzt
FR - Pour usage professionnel seulement
NL - Alleen voor professioneel gebruik
IT - Solo per uso professionale
ES - Solo para uso profesional
PT - Apenas para utilização profissional
SV - Endast för professionell användning
DA - Kun til professionel brug
FI - Vain ammattikäyttöön
EL - Για επαγγελματική χρήση μόνοNO - Kun for profesjonell brukPL - Wyrób wyłącznie do użytku profesjonalnego.
RU - Только для профессионального использования
TR - Sadece profesyonel kullanım için
AR - ‫لالستخدام من قِبل متخصصين فقط‬TH - สำำ�หรัับการใช้้งานระดัับมืืออาชีีพเท่่านั้้�น
KO - 전문가 전용
ZH - 仅用于专业用途

EN - Catalog Number
DE - Katalognummer
FR - Numéro de catalogue
NL - Catalogusnummer
IT - Numero di catalogo
ES - Numero de catálogo
PT - Número de catálogo
SV - Katalognummer
DA - Katalognummer
FI - Tuotenumero
EL - Αριθμός καταλόγου
NO - Katalognummer
PL - Numer katalogowy
RU - Номер каталога
TR - Katalog Numarası
AR - ‫رقم الفهرس‬TH - หมายเลขแคตาล็็อก
KO - 카탈로그 번호
ZH - 目录号Flammability

1

0

1

Reactivityรายละเอียด:‫الوصف‬Ultra-Blend™ เป็ นวัสดุรองพื้นและฉาบที่ยึดติดกับเดนทีน บ่มตัวด้วยแสง และทึบรังสี ซึ่งผสม
‫ يتم تصليبه بالضوء وهو‬،‫ هو مبطن يلتصق بالعاج ومادة قاعدية‬Ultra-Blend™ plusแคลเซียมไฮดร็อกไซด์และแคลเซียมไฮดรอกซีแอพพา ไทต์ในยูรเี ธนไดเมธาไคลเลทเบส (UDMA)‫ ويحتوي عىل هيدروكسيد الكالسيوم وهيدروكسي أباتيت الكالسيوم‬،‫ظليل لألشعة‬่่
Ultra-Blend plus มีปริมาณสารเติมระดับสูงเพือลดการหดตัวให้เหลือน้อยทีสุดและมีความหนืด
‫ به كمية كبيرة من‬UDMA). Ultra-Blend plus( ‫في قاعدة يوريثين ثنائي ميثا أكريالت‬สูงเพื่อช่วยให้มกี ารทาฉาบได้ง่าย สามารถจัดรูปร่างวัสดุได้ด้วยด้ามกรอความเร็วสูงหรือต่ำ� และมี
‫ يمكن تكييف حوافه‬.‫ ولزوجة عالية لتسهيل عملية وضعه‬،‫الحشو للحد من االنكماش‬มาในเฉดสีแบบเดนทีนและสีขาวทึบแสง
‫ ويتوفر بألوان العاج واأللوان‬،‫بسهولة باستخدام جهاز يدوي ذو سرعة عالية أو منخفضة‬.‫البيضاء المعتمة‬ข้อบ่งใช้:่้
:‫المؤشرات‬UltraBlend plus เป็ นวัสดุทียึดติดได้ด้วยตัวเองและใช้เป็ นสารรองพืนโพรงฟั น และ/หรือ วัสดุ‫أو كمادة قاعدية قبل‬/‫ يلتصق ذاتيا ويستخدم كمبطن للحفر السنية و‬Ultra-Blend plusฉาบก่อนการใส่วสั ดุบูรณะฟั น.‫وضع المواد الترميمية‬ี ารใช้:วิธก:‫إرشادات االستخدام‬A. ทั่วไป
‫ عام‬-‫أ‬
1. ติดปลาย Black Micro™ กับหลอดฉีดให้แน่น สำ�หรับการเตรียมฟั นขนาดใหญ่ ให้ใช้
‫ استخدم‬،‫ للتحضيرات األكبر حجما‬.‫™ بأمان في المحقنة‬Black Micro ‫ قم بتثبيت طرف‬- 1ปลาย Black Mini™
™
. Black Mini ‫طرف‬
2. เตรียมบริเวณด้วยระบบการยึดที่มีคณุ ภาพตามขัน้ ตอนของผู้ผลิต
.‫ قم بإعداد المنطقة باستخدام نظام لصق عال الجودة وفقا لتعليمات الشركة المصنعة‬- 2
3. ทาชัน้ Ultra-Blend plus หนา .5-2 มม. บนบริเวณที่ต้องการ
.‫ مم في المنطقة المطلوبة‬2 – 0,5 ‫ بسمك‬Ultra-Blend plus ‫ ضع طبقة من‬- 3หมายเหตุ: เมื่อทาบนบริเวณที่เข้าถึงได้ยาก ให้บบี วัสดุปริมาณเล็กน้อยออกมาเพื่อให้ก่อรูปเป็ น‫ قم بضخ كمية صغيرة لتكون‬،‫ عند وضعه في منطقة يصعب الوصول إليها‬:‫مالحظة‬ลูกบอลกลมที่ปลาย ใช้ปลายเป็ นแปรงปลายลูกบอลกลมเพื่อป้ องกันการบีบออกมามากเกินไป
.‫ استخدم الطرف لوضع الكرة لمنع اإلفراط في التصريف‬.‫كرة صغيرة عند الطرف‬4. บ่มตัวด้วยการฉายแสงด้วยเครื่องฉายแสง VALO™ LED Curing Light ด้วยระดับ
‫ باستخدام طاقة قياسية لمدة‬VALO™ LED ‫ قم بالتصليب باستخدام ضوء التصليب‬- 4Standard Power เป็ นเวลา 10 วินาที (สำ�หรับแสงที่มีกำ�ลังไฟ <600mW/cm2 ให้บม่เป็ นเวลา 20 วินาที หรือสำ�หรับแสงที่มีกำ�ลังไฟ >600mw/cm2 ให้บม่ เป็ นเวลา 10 วินาที) ‫ أو‬،‫ ثانية‬20 ‫ قم بالتصليب لمدة‬2‫ سم‬/ ‫ م و‬600 < ‫ ثواني (لألضواء التي لها إخراج‬10.)‫ ثواني‬10 ‫ قم بالتصليب لمدة‬2‫ سم‬/ ‫ م و‬600 > ‫لألضواء التي لها إخراج‬5. หมายเหตุ: สำ�หรับการสร้างชัน้ ฉาบ ให้เริม่ ต้นด้วยชัน้ ความหนา 0.5 มม. แล้วฉายแสงบ่ม
‫ اتبع‬.‫ مم وقم بتصليبها بالضوء‬0,5 ‫ ابدأ بوضع طبقة سمكها‬،‫ لبناء القاعدة‬:‫مالحظة‬ตามด้วยชัน้ หนา 2 มม. แล้วบ่มระหว่างการทาแต่ละชัน้
.‫ وقم بتصليب كل طبقة‬،‫ مم‬2 ‫بعد ذلك بطبقات سمكها‬6. Ultra-Blend plus จะยึดติดกับพื้นผิวที่เตรียมไว้ด้วยสารยึดติดเดนทีนและเพื่อทำ�ให้
‫لصق‬‫عوامل‬‫ باألسطح التي تم إعدادها باستخدام‬Ultra-Blend plus ‫ سوف يلتصق‬- 5เดนทีนแห้ง
.‫ وللعاج الجاف‬،‫العاج‬B. สารรองพื้นเดนทีน
‫ مبطن العاج‬-‫ب‬
1. ล้างทำ�ความสะอาดเดนทีนที่ถูกเปิ ดออกด้วยน้ำ�แล้วเป่ าลมให้แห้ง
.‫ اشطف العاج المكشوف بالماء وجففه بالهواء‬- 1
2. ทาด้วยระบบการยึดที่มีคณุ ภาพตามขัน้ ตอนของผู้ผลิต่.‫ ضع نظام لصق عال الجودة وفقا لتعليمات الشركة المصنعة‬- 2
3. ทา Ultra-Blend plus อย่างแม่นยำ�และในปริมาณทีเท่ากันตามต้องการด้วยปลาย
Black Micro
Black ‫ بدقة وبالحجم الصحيح المطلوب باستخدام طرف‬Ultra-Blend plus ‫ ضع‬- 3
4. บ่มตัวด้วยการฉายแสงด้วยเครื่องฉายแสง VALO™ LED Curing Light ด้วยระดับ
.Micro‫ القاعدة الواقية‬-‫ج‬Standard Power เป็ นเวลา 10 วินาที (สำ�หรับแสงที่มีกำ�ลังไฟ <600mW/cm2 ให้บ่มเป็ นเวลา 20 วินาที หรือสำ�หรับ แสงที มีกำ�ลังไฟ >600mw/cm2 ให้บม่ เป็ นเวลา
.‫ اشطف العاج المكشوف بالماء وجففه بالهواء‬- 1
10 วินาที).‫ ضع نظام لصق عال الجودة وفقا لتعليمات الشركة المصنعة‬- 2
C. สารฉาบเพื่อป้ องกัน
‫ استخدم‬،‫ في المحقنة إذا كانت المنطقة األساسية صغيرة‬Black Micro ‫ ضع طرف‬- 3
1. ล้างทำ�ความสะอาดเดนทีนที่ถูกเปิ ดออกด้วยน้ำ�แล้วเบ่าลมให้แห้ง
.‫ للمناطق الكبيرة‬Black Mini ‫طرف‬่้
2. ทาด้วยระบบการยึดทีมีคณุ ภาพตามขันตอนของผู้ผลิต
‫ باستخدام‬VALO™ LED ‫ وقم بتصليبه باستخدام ضوء التصليب‬Ultra-Blend plus ‫ ضع‬- 4
3. ติดปลาย Black Micro เข้ากับหลอดฉีดหากบริเวณฉาบมีขนาดเล็ก ให้ใช้ปลาย Black‫ قم‬2‫ سم‬/ ‫ م و‬600 < ‫ ثواني (لألضواء التي لها إخراج‬10 ‫طاقة قياسية لمدة‬Mini สำ�หรับบริเวณที่ใหญ่กว่า
‫بالتصليب‬
‫قم‬
2‫سم‬
/
‫و‬
‫م‬
600
>
‫إخراج‬‫لها‬‫التي‬‫لألضواء‬‫ أو‬،‫ ثانية‬20 ‫بالتصليب لمدة‬4. ทา Ultra-Blend plus แล้วบ่มตัวด้วยการฉายแสงด้วยเครื่องฉายแสง VALO™ LED.)‫ ثواني‬10 ‫لمدة‬Curing Light ด้วยระดับ Standard Power เป็ นเวลา 10 วินาที (สำ�หรับแสงที่มีกำ�ลัง
‫ ضع طبقة أوىل بسمك‬،‫ مم‬2 ‫ إذا كانت الطبقة النهائية يجب أن تكون بسمك أكبر من‬- 5
ไฟ <600mW/cm2 ให้บม่ เป็ นเวลา 20 วินาที หรือสำ�หรับแสงที่มีกำ�ลังไฟ >600mw/‫ ضع طبقات إضافية بسمك‬.‫ ثواني‬10 ‫ مم (أو أقل) وقم بتصليبها بالضوء لمدة‬0,5
cm2 ให้บม่ เป็ นเวลา 10 วินาที).‫ مم حسب الحاجة‬2
5. หากการฉาบขัน้ สุดท้ายต้องมีความหนามากกว่า 2 มม. ให้ทาชัน้ แรกที่ความหนา 0.5.‫ مم‬3 ‫ بالضوء لعمق‬Ultra-Blend plus ‫ يمكن تصليب‬:‫مالحظة‬มม. (หรือน้อยกว่านัน้ ) แล้วบ่มด้วยแสงเป็ นเวลา 10 วินาที เพิม่ ชัน้ ต่อๆ ไปที่ความหนา
‫ تغطية اللب‬-‫د‬
2 มม. ตามที่ต้องการ
‫ بالقرب من اللب (وردي) والتعرض الصغير‬Ultra-Blend plus ‫ استخدم‬:‫مالحظة‬‫ يجب البدء‬،‫أو إذا كان اللب تبيغي‬/‫ و‬،‫ إذا كان التعرض كبير‬.‫الميكانيكي الغير تبيغي‬หมายเหตุ: Ultra-Blend plus จะบ่มตัวด้วยแสงที่ความลึก 3 มม.
.‫في معالجة اللب‬D. การปิ ดทับเนื้อเยื่อใน
™
หมายเหตุ: ใช้ Ultra-Blend plus กับส่วนที่ใกล้เนื้อเยื่อใน (สีชมพู ) และการเผยเชิงกลที่ไม่
Mini Dento-Infusor ‫™ المضاد للجراثيم باستخدام طرف‬Consepsis ‫ ضع محلول‬- 1.‫ جفف بالهواء بلطف‬.‫ ال تشطف‬.‫ ثانية باستخدام حك خفيف‬60 ‫بالستيك لمدة‬อักเสบขนาดเล็ก หากการเผยขนาดใหญ่กว่า และ/หรือ มีอาการอักเสบของเนื้อเยื่อใน ควรเริม่
ทำ�การรักษาโรคเนื้อเยื่อในฟั น‫ عىل العاج الجاف الموجود‬Black Micro ‫ باستخدام طرف‬Ultra-Blend plus ‫ ضع‬- 2
1. ทาละลายป้ องกันแบคทีเรีย Consepsis™ ด้วยปลายพลาสติก Mini Dento-Infusor™
‫ أو فوق العاج الوردي الرفيع للتعرض‬،‫ ثم بعد ذلك فوق التعرض‬،‫في حدود التعرض‬.‫القريب‬เป็ นเวลา 60 วินาที โดยการถูเบาๆ ห้ามล้างทำ�ความสะอาด ค่อยๆ เป่ าลมให้แห้ง
.‫ قم بتقليل تغطية العاج لزيادة المساحة المتاحة لتقنيات لصق العاج‬:‫مالحظة‬2. ทา Ultra-Blend plus ด้วยปลาย Black Micro ลงบนเดนทีนที่แห้งแล้วที่ขอบของการเผย จากนัน้ ทาสารบนการเผยหรือบนเดนทีนสีชมพู ของการเผยที่อยูใ่ กล้‫ باستخدام طاقة قياسية لمدة‬VALO™ LED ‫ قم بالتصليب باستخدام ضوء التصليب‬- 3‫ أو‬،‫ ثانية‬20 ‫ قم بالتصليب لمدة‬2‫ سم‬/ ‫ م و‬600 < ‫ ثواني (لألضواء التي لها إخراج‬10หมายเหตุ: ลดการทาบนเดนทีนให้น้อยที่สุดเพื่อให้มบี ริเวณเหลือมากที่สุดสำ�หรับเทคนิค
.)‫ ثواني‬10 ‫ قم بالتصليب لمدة‬2‫ سم‬/ ‫ م و‬600 > ‫لألضواء التي لها إخراج‬การยึดเดนทีน
3. บ่มตัวด้วยการฉายแสงด้วยเครื่องฉายแสง VALO™ LED Curing Light ด้วยระดับ
‫ وقم بالترميم بالمنتج‬،‫ ضع نظام لصق عال الجودة وفقا لتعليمات الشركة المصنعة‬- 4.‫الذي تختاره‬Standard Power เป็ นเวลา 10 วินาที (สำ�หรับแสงที่มีกำ�ลังไฟ <600mW/cm2 ให้บม่
‫ المعتم‬-‫ﮪ‬เป็ นเวลา 20 วินาที หรือสำ�หรับแสงที่มีกำ�ลังไฟ >600mw/cm2 ให้บม่ เป็ นเวลา 10 วินาที)4. ทาระบบการยึดที่มีคณุ ภาพตามขัน้ ตอนของผู้ผลิต และบูรณะฟั นด้วยผลิตภัณฑ์ที่
.‫ قم بتخشين المعدن باستخدام مثقب أو بالسحج المكروي‬-- 1ท่านต้องการใช้
.‫ اشطف المعدن المكشوف الذي سوف يتم تعقيمه بالماء وجففه بالهواء‬- 2
E. สารทึบแสง
‫ وقم بالتصليب‬،‫ ضع نظام لصق عال الجودة وفقا لتعليمات الشركة المصنعة‬- 3
1. ทำ�ให้โลหะหยาบด้วยหัวกรอหรือการขัดแบบเล็ก
.‫باستخدام الضوء‬2. ล้างทำ�ความสะอาดโลหะที่มองเห็นที่จะทำ�ให้ทึบแสงด้วยน้ำ�แล้วเป่ าลมให้แห้ง
.‫ مم‬2 ‫ فوق المعدن تدريجيا بحيث ال يزيد سمك أي طبقة عن‬Ultra-Blend plus ‫ ضع‬- 4
3. ทาด้วยระบบการยึดที่มีคณุ ภาพตามขัน้ ตอนของผู้ผลิตแล้วบ่มด้วยแสง
‫ باستخدام طاقة قياسية لمدة‬VALO™ LED ‫قم بالتصليب باستخدام ضوء التصليب‬4. ทาชัน้ Ultra-Blend plus บนโลหะโดยทาเพิม่ ขึ้นเป็ นชัน้ ๆ โดยจะไม่มชี นั้ ที่หนาเกิน 2‫ أو‬،‫ ثانية‬20 ‫ قم بالتصليب لمدة‬2‫ سم‬/ ‫ م و‬600 < ‫ ثواني (لألضواء التي لها إخراج‬10
มม. บ่มตัวด้วยการฉายแสงด้วยเครื่องฉายแสง VALO™ LED Curing Light ด้วยระดับ
.)‫ ثواني‬10 ‫ قم بالتصليب لمدة‬2‫ سم‬/ ‫ م و‬600 > ‫لألضواء التي لها إخراج‬Standard Power เป็ นเวลา 10 วินาที (สำ�หรับแสงที่มีกำ�ลังไฟ <600mW/cm2 ให้บม่
.‫ قم بالترميم بالمنتج الذي تختاره‬- 5
เป็ นเวลา 20 วินาที หรือสำ�หรับแสงที่มีกำ�ลังไฟ >600mw/cm2 ให้บม่ เป็ นเวลา 10 วินาที):‫االحتياطات‬5. บูรณะด้วยคอมโพสิทหรือผลิตภัณฑ์ที่ท่านต้องการเลือกใช้،‫ لتجنب االنسداد‬.‫ قد تتصلب المادة الكيميائية في الطرف بعد عدة ساعات قليلة‬:‫تحذير‬ข้อควรระวัง:‫ تحقق دائما من تدفق‬.‫ بين كل استعمال‬Luer Lock ‫أزل الطرف وأعد تغطيتها بأغطية‬่่่คำ�เตือน: สารเคมีอาจแข็งตัวในปลายหลังจากเวลาผ่านไปสองสามชัวโมง เพือหลีกเลียงการอุ ด.‫ إذا قابلت مقاومة قم باستبدال الطرف وأعد االختبار‬.‫المواد قبل وضعها داخل الفم‬.‫استخدم األطراف الموصي بها فقط‬ตัน ให้นำ�ปลายออกและปิ ดฝาครอบด้วยฝา Luer Lock Caps ระหว่างการใช้งานแต่ละครัง้ ตรวจ่สอบการไหลก่อนที่จะใชงานภายในปากเสมอ หากพบว่าสารไหลได้ยาก ให้เปลียนปลายแล้วตรวจ‫ اقرأ وتفهم جميع التعليمات بعناية وصحيفة بيانات سالمة المواد قبل استخدام‬- 1สอบใหม่ ใช้ปลายที่แนะนำ�ให้ใช้เท่านัน้
.‫ استخدمه وفقا للتوجيهات فقط‬.Ultra-Blend plus้
1. อ่านและทำ�ความเข้าใจ SDS และขันตอนต่างๆ อย่างรอบคอบก่อนใช้ Ultra-Blend plus‫ يجب أن يرتدي الطبيب والمريض نظارات واقية من األشعة فوق البنفسجية عند‬- 2ใช้งานตามที่ระบุไว้เท่านัน้
.Ultra-Blend plus ‫تصليب‬2. แพทย์ผู้รกั ษาและผู้ป่วยควรสวมใส่แว่นป้ องกันแสง UV เมื่อฉายแสงบ่มตัว Ultra-Blend.‫ قم بتبريد الكميات الكبيرة‬.‫ قم بتخزينه في درجة حرارة الغرفة‬،‫ لالستخدام اليومي‬- 3plus.‫ قم بإعادة توجيه الضوء العلوي لمنع البلمرة المسبقة‬- 4
่
3. สำ�หรับการใช้งานประจำ�วัน ให้เก็บวัสดุไว้ทีอุ ณหภูมหิ อ้ ง แช่เย็นวัสดุปริมาณมาก
‫ تجنب التالمس المتكرر لراتين األسنان الغير‬.‫ يمكن أن تسبب الراتينات حساسية‬- 5
4. หันโคมไฟส่องปากไปทางอื่นเพื่อป้ องกันการทำ�ปฏิกิรยิ ากับแสงก่อนเวลาสมควร
‫ ال تستخدمه أثناء عالج المرضى الذين يعانون من حساسية‬.‫متصلب مع الجلد‬่
5. ท่านสามารถฆ่าเชื้อกับเรซินได้ หลีกเลียงไม่
ให้เรซินทันตกรรมที่ยังไม่บม่ ตัวถูกกับผิวหนังบ่อย،‫ أو التهاب للجلد‬،‫ إذا حدث رد فعل تحسسي‬.‫معروفة لألكريالت أو الراتينات األخرى‬ครัง้ อย่าใช้กับผู้ป่วยที่มีความไวต่ออะไครเลทหรือเรซินชนิดอื่นๆ หากมีอาการแพ้ ผิวหนัง
.‫ استشر طبيب‬،‫أو طفح‬่้อักเสบ หรือผืนคันเกิดขึน ให้ปรึกษาแพทย์
.‫ تخلص من األطراف المستعملة والمحاقن الفارغة بطريقة مناسبة‬- 6
6. ทิ้งปลายและหลอดเปล่าที่ใช้แล้วอย่างถูกต้องเหมาะสม
.‫ احفظه بعيدا عن متناول األطفال‬- 7
7. เก็บให้หา่ งจากมือเด็ก
.‫ ال تقم بإعادة استخدام األطراف‬،‫ لتجنب التلوث المتبادل‬- 8่้ อนข้าม อย่านำ�ปลายกลับมาใช้ใหม่อีก
8. เพื่อหลีกเลียงการปนเปื
.‫ ا تستخدمه بعد تاريخ انتهاء الصالحية المكتوب عىل الحاويات‬- 9
9. ห้ามใช้หลังวันที่หมดอายุที่ระบุไว้บนบรรจุภัณฑ์
่้ อนข้าม ให้ฆา่ เชื้อและเช็ดหลอดฉีดด้วยยาฆ่าเชื้อประสิทธิภาพปาน
‫ قم بتعقيم ومسح المحاقن باستخدام مطهر متوسط‬،‫ لتجنب التلوث المتبادل‬-10	
10. เพื่อหลีกเลียงการการปนเปื
‫ إذا تم‬.‫أو استخدم غطاء محقنة يستخدم مرة واحدة‬/‫المستوى بين االستخدامات و‬กลางระหว่างการใช้งาน และ/หรือ ใช้ฝาครอบหลอดฉีดแบบใช้ครัง้ เดียว หากฝาครอบหลอด
.‫ وتخلص من الطرف وغطاء المحقنة‬،‫ أعد التغطية‬،‫ أزل الطرف‬،‫استخدام غطاء محقنة‬ฉีดถูกใช้แล้ว ให้ถอดปลายออก ปิ ดฝา แล้วทิ้งปลายและฝาครอบหลอดฉีด
.‫ بعيدا عن الحرارة وضوء الشمس المباشر‬Ultra-Blend plus ‫ احفظ‬-11
11. เก็บ Ultra-Blend plus ให้หา่ งจากความร้อน/แสงอาทิตย์โดยตรงTR

HAZARD RATING
4 = Severe
3 = Serious
2 = Moderate
1 = Slight
0 = Minimal

EN - Use by date
DE - Verfallsdatum
FR - Date de péremption
NL - Uiterste gebruiksdatum
IT - Utilizzare entro
ES - Utilizar antes de la fecha
PT - Utilizar antes deSV - Använd före
DA - Sidste anvendelsesdato
FI - Viimeinen käyttöpäivä
EL - Ημερομηνία λήξης
NO - Utløpsdato
PL - Data ważności
RU - Использовать до
TR - Son kullanma tarihi
AR - ‫تاريخ انتهاء الصالحية‬TH - ใช้้ก่อ่ นวัันที่่�
KO - 사용 기한
ZH - 有效期EN - Do not re-use to avoid cross contamination
DE - Nicht wiederverwenden, um Kreuzkontamination zu vermeiden
FR - Ne pas réutiliser pour éviter la contamination croisée
NL - Niet opnieuw gebruiken om kruisbesmetting te voorkomen
IT - Non riutilizzare per prevenire la contaminazione crociata
ES - No reutilizar para evitar la contaminación cruzada.
PT - Não reutilize para evitar contaminação cruzada
SV - Återanvänd inte för undvikande av korskontamination
DA - Må ikke genanvendes for at undgå krydskontaminering
FI - Älä käytä uudelleen ristikontaminaation välttämiseksi
EL - Μην επαναχρησιμοποιείτε για να αποφύγετε τη διασταυρούμενη μόλυνσηNO - Skal ikke gjenbrukes for å unngå krysskontaminering
PL - Nie używaj ponownie, aby uniknąć skażenia krzyżowego
RU - Не используйте повторно во избежание перекрестного заражения
TR - Çapraz kontaminasyonu önlemek için tekrar kullanmayın
AR - ‫يُحظر إعادة استخدامه لتالفي انتشار التلوث‬TH - อย่่าใช้้ซ้ำ��ำ เพื่่� อหลีีกเลี่่�ยงการปนเปื้้� อนข้้าม
KO - 교차 오염 방지를 위해 재사용하지 말 것
ZH - 切勿再使用，以免交叉感染

EN - Destroy after use
DE - Nach Gebrauch vernichten
FR - Détruire après utilisation
NL - Vernietig na gebruik
IT - Distruggere dopo l'uso
ES - Destruir después de usar
PT - Destruir após utilização
SV - Förstör efter användning
DA - Destrueres efter brug
FI - Tuhoa käytön jälkeen
EL - Καταστρέψτε μετά τη χρήση
NO - Destruer etter brukPL - Należy zniszczyć po użyciu
RU - Уничтожить после использования
TR - Kullandıktan sonra imha edin
AR - ‫اإلتالف بعد االستخدام‬TH - ทำำ�ลายหลัังการใช้้งาน
KO - 사용 후 파기
ZH - 使用后销毁

EN - See Instructions
DE - Siehe Anleitung
FR - Voir les instructions
NL - Zie instructies
IT - Vedere le istruzioni
ES - Consulte las instrucciones
PT - Veja as instruções
SV - Se anvisningar
DA - Se vejledning
FI - Katso ohjeet
EL - Βλέπε Οδηγίες
NO - Se instruksjonene
PL - Zobacz Instrukcje
RU - См. инструкции
TR - Talimatlara Bakın
AR - ‫انظر اإلرشادات‬TH - ดููคำำ�แนะนำำ�ในการใช้้งาน
KO - 설명서 참고
ZH - 参阅说明书EN - Not for injection
DE - Nicht zur Injektion verwenden
FR - N'est pas fait pour être injecté
NL - Niet voor injectie
IT - Non iniettabile
ES - No inyectable
PT - Não deve ser injetado
SV - Ej för injektion
DA - Ikke til injektion
FI - Ei injektiona
EL - Όχι για ένεση
NO - Ikke til injeksjon
PL - Wyrób nie jest przeznaczony dowstrzykiwań
RU - Не для инъекций
TR - Enjekte edilmez
AR - ‫غير مخصص للحقن‬TH - ไม่่ได้้มีไี ว้้สำ�ำ หรัับการฉีีด
KO - 주사용 아님
ZH - 不用于注射EN - Recycle
DE - Wiederaufbereiten
FR - Recycler
NL - Recyclen
IT - Riciclare
ES - Reciclar
PT - Reciclar
SV - Återvinn
DA - Genbrug
FI - Kierrätys
EL - Ανακυκλώνετε
NO - Resirkuleres
PL - Utylizacja
RU - Утилизировать
TR - Geri dönüştürün
AR - ‫إعادة التدوير‬TH - รีีไซเคิิล
KO - 재활용
ZH -回收

2°C
36°F

27°C
81°F

41567-UAR11 062121

TH

Ultra-Blend™ Plus

Kalsiyum Hidroksit Dentin Liner ve Koruyucu Baz

Tanım:
Ultra-Blend™ plus ışıkla sertleşen, radyopak bir dentin yapışkan liner ve üretan
dimetakrilat (UDMA) baz içinde kalsiyum hidroksit ve kalsiyum hidroksiapatit içeren
bir baz materyaldir. Ultra-Blend plus, büzülmeyi minimize etmek için yüksek dolgu
yüküne ve yerleştirmeyi kolaylaştırmak için yüksek viskoziteye sahiptir. Yüksek veya
düşük hızlı el cihazı ile kolayca şekillendirilebilir ve dentin ve opak beyaz tonları
mevcuttur.
Endikasyonlar:
UltraBlend plus, kendinden yapışkanlıdır ve restoratif materyaller yerleştirilmedenönce kavite liner ve/veya baz materyal olarak kullanılır.
Kullanım Talimatları:
A. Genel
1. Black Micro™ ucu şırıngaya sıkıca takın. Daha büyük preparatlar için Black
Mini™ uç kullanın.
2. Alanı üreticinin talimatlarına göre kaliteli bir bonding sistemiyle hazırlayın.
3. İstenilen alana 0,5 – 2 mm Ultra-Blend plus katmanı uygulayın.
NOT: Erişilmesi güç olan alanlara uygularken uçta bir top oluşturmak için
küçük bir miktar uygulayın. Aşırı damlatmamak için ucu bir top uygulayıcı
olarak kullanın.
4. VALO™ LED Sertleştirme Işığı ile Standard Güçte 10 saniye boyunca ışıkla
sertleştirin (çıkış gücü <600mW/cm2 olan ışıklar için 20 saniye veya >600mw/
cm2 olan ışıklar için 10 saniye boyunca sertleştirin).
NOT: Baz oluşturmak için 0,5 mm’lik katmanla başlayın ve ışıkla sertleştirin. Her
inkrementi polimerize ederek 2 mm’lik inkrementlerle devam edin.
5. Ultra-Blend plus, dentin bonding ajanları ile hazırlanmış yüzeylere ve kuru
dentine yapışacaktır.
B. Dentin Liner
1. Ekspoze olmuş dentini suyla yıkayın ve kurumaya bırakın.
2. Üreticinin talimatlarına göre kaliteli bir bonding sistemi uygulayın.
3. Ultra-Blend plus’ı Black Micro uçla doğru bir şekilde ve tam olarak istenilen
miktarda yerleştirin.
4. VALO™ LED Sertleştirme Işığı ile Standard Güçte 10 saniye boyunca ışıkla
sertleştirin (çıkış gücü <600mW/cm2 olan ışıklar için 20 saniye veya >600mw/
cm2 olan ışıklar için 10 saniye boyunca sertleştirin).
C. Koruyucu Baz
1. Ekspoze olmuş dentini suyla yıkayın ve kurumaya bırakın.
2. Üreticinin talimatlarına göre kaliteli bir bonding sistemi uygulayın.
3. Baz alanı küçükse şırıngaya Black Micro uç takın; daha büyük alanlar için Black
Mini uç kullanın.
4. Ultra-Blend plus uygulayın ve VALO™ LED Sertleştirme Işığı ile Standard Güçte
10 saniye boyunca ışıkla sertleştirin (çıkış gücü <600mW/cm2 olan ışıklar için
20 saniye veya >600mw/cm2 olan ışıklar için 10 saniye boyunca sertleştirin).
5. Final bazının 2 mm’den kalın olması gerekiyorsa ilk katman olarak 0,5 mm(veya daha az) uygulayın ve 10 saniye boyunca ışıkla sertleştirin. Gerektiğinde
2 mm kalınlığında ilave katman ekleyin.
Not: Ultra-Blend plus 3 mm derinliğe kadar ışıkla sertleştirilebilir.
D. Pulpa Kaplama
NOT: Pulpanın (pembe) yanındaki ve hiperemik olmayan küçük mekanik ekspozüriçin Ultra-Blend plus kullanın. Daha büyük ekspozür ve/veya hiperemik pulpa içinendodontik terapi başlatılmalıdır.
1. Plastik Mini Dento-Infusor™ uçla nazik bir şekilde ovarak 60 saniye boyunca
Consepsis™ antibakteriyel solüsyon uygulayın. Yıkamayın. Nazik bir şekilde
kurumaya bırakın.
2. Ekspozürün kenarındaki kuru dentine, daha sonra ekspozüre veya ekspozürün
yanındaki ince pembe dentine Black Micro uçla Ultra-Blend plus uygulayın.
NOT: Dentin bonding teknikleri için kullanılabilen alanı maksimuma çıkarmak
amacıyla dentini mümkün olduğunca az kapatın.
3. VALO™ LED Sertleştirme Işığı ile Standard Güçte 10 saniye boyunca ışıkla
sertleştirin (çıkış gücü <600mW/cm2 olan ışıklar için 20 saniye veya >600mw/
cm2 olan ışıklar için 10 saniye boyunca sertleştirin).
4. Üreticinin talimatlarına göre kaliteli bir bonding sistemi uygulayın ve seçtiğiniz
ürünle restore edin.
E. Opaklaştırıcı
1. Bur veya mikroabrazyon ile metali pürüzlendirin.
2. Ekspoze olmuş opaklanacak metali suyla yıkayın ve kurumaya bırakın.
3. Üreticinin talimatlarına göre kaliteli bir bonding sistemi uygulayın ve ışıkla
sertleştirin.
4. Katmanlardan hiçbiri 2 mm’den kalın olmayacak şekilde metalin üzerine
inkremental olarak Ultra-Blend plus uygulayın. VALO™ LED Sertleştirme Işığı
ile Standard Güçte 10 saniye boyunca ışıkla sertleştirin (çıkış gücü <600mW/
cm2 olan ışıklar için 20 saniye veya >600mw/cm2 olan ışıklar için 10 saniye
boyunca sertleştirin).
5. Seçtiğiniz kompozit veya ürünle restore edin.
Önlemler:
UYARI: Kimyasal birkaç saat sonra uçta donabilir. Tıkanmayı önlemek için her
kullanımdan sonra ucu çıkarıp Luer Lock Kapak takın. Ağız içine uygulamadan öncemutlaka materyalin
akıp akmadığını kontrol edin. Dirençle karşılaşılırsa ucu değiştirin ve yeniden
kontrol edin. Sadece önerilen uçları kullanın.
1. Ultra Blend plus’ı kullanmadan önce tüm talimatları ve Malzeme Güvenlik
Formu’nu dikkatli bir şekilde okuyun ve anlayın. Sadece talimatlar doğrultusunda
kullanın.
2. Ultra-Blend plus polimerize edilirken klinisyen ve hasta UV koruyucu gözlük
takmalıdır.
3. Günlük kullanım için oda sıcaklığında muhafaza edin. Toplu depolamalardasoğutucuda saklayın.
4. Erken polimerizasyonu önlemek için tepe ışığını başka yöne çevirin.
5. Rezinler duyarlılık oluşturabilir. Sertleştirme işlemi yapılmamış dental rezinin
deriyle tekrarlanan temasından kaçının. Akrilat veya diğer rezinlere karşı
duyarlılığı olduğu bilinen hastalarda kullanmayın. Alerjik reaksiyon, dermatit veya
deri döküntüsü oluşursa bir hekime danışın.
6. Kullanılan uçları ve boş şırıngaları uygun biçimde elden çıkarın.
7. Çocukların ulaşamayacağı yerlerde muhafaza edin.
8. Çapraz kontaminasyonu önlemek için uçları tekrar kullanmayın.
9. Kapların üzerinde belirtilen son kullanma tarihi geçtikten sonra kullanmayın.
10. Çapraz kontaminasyonu önlemek için kullanımlar arasında şırıngayı orta düzeyli
bir dezenfektanla dezenfekte edip silin ve/veya tek kullanımlık şırınga kapağı
kullanın. Şırınga kapağı kullanılırsa ucu çıkarın, kapağı takın, ucu ve şırıngakapağını atın.
11. Ultra-Blend plus’ı sıcaktan/doğrudan güneş ışından uzak tutun.

u l t r a d e n t . c o m

Ultra-Blend™ Plusสารรองพื้นและสารฉาบเพื่อป้ องกันเดนทีนที่ผสมแคลเซียมไฮดร็อกไซด์

PL

|

ZH

Ultra-Blend™ PlusPodścielacz i podkład ochronny z wodorotlenkiem wapnia

Opis:
Preparat Ultra-Blend™ plus to światłoutwardzalny, nieprzepuszczalny dla promieni
rentgenowskich, przywierający do zębiny podścielacz i podkład zawierający wodorotlenek
wapnia i hydroksyapatyt wapnia w podłożu z dimetakrylanu uretanu (UDMA). Preparat
Ultra-Blend plus ma wysoką zawartość wypełniacza, aby zminimalizować kurczenie, iwiększą lepkość, aby ułatwić wprowadzanie. Może być łatwo konturowany za pomocą
końcówek stomatologicznych o dużej lub małej prędkości i jest dostępny w odcieniachzębiny i nieprzezroczystym białym.
Wskazania:
Ultra-Blend plus jest preparatem samoprzylepnym i stosowanym jak podścielacz ubytków
i/lub podkład, przed umieszczeniem materiałów do wypełnień.
Sposób użycia:
A. Ogólne
1. Pewnie zamocować końcówkę Black Micro™ do strzykawki. W przypadku
opracowywania większych obszarów należy użyć końcówki Black Mini™.
2. Przygotować obszar przy pomocy wysokiej jakości systemu wiążącego, zgodnie z
instrukcją producenta.
3. W żądanym obszarze nałożyć warstwę preparatu Ultra-Blend plus o grubości
0,5-2 mm.
UWAGA: podczas nakładania w obszarze trudnodostępnym należy wycisnąć niewielką
ilość preparatu, aby na końcówce uformować kulkę. Użyć końcówki jako aplikatora
kulki, aby zapobiec wyciśnięciu nadmiernej ilości.
4. Przez 10 sekund utwardzać za pomocą lampy polimeryzacyjnej VALO™ LED (wprzypadku lamp o mocy <600 mW/cm2 utwardzać przez 20 sekund, a w przypadku
lamp o mocy >600 mW/cm2 - przez 10 sekund).
UWAGA: w celu nagromadzenia podkładu należy rozpocząć od warstwy o grubości 0,5mm i utwardzić. Następnie dokładać po 2 mm i utwardzać po każdej warstwie.
5. Preparat Ultra-Blend plus wiąże się z powierzchniami opracowywanymi za pomocą
środków wiążących zębinę i z suchą zębiną.
B. Podścielacz zębinowy
1. Odsłoniętą zębinę przepłukać wodą i osuszyć dmuchawką.
2. Nałożyć wysokiej jakości system wiążący, zgodnie z instrukcją producenta.
3. Za pomocą końcówki Black Micro dokładnie nałożyć żądaną objętość preparatu
Ultra-Blend plus.
4. Przez 10 sekund utwardzać za pomocą lampy polimeryzacyjnej VALO™ LED (wprzypadku lamp o mocy <600 mW/cm2 utwardzać przez 20 sekund, a w przypadku
lamp o mocy >600 mW/cm2 - przez 10 sekund).
C. Podkład ochronny
1. Odsłoniętą zębinę przepłukać wodą i osuszyć dmuchawką.
2. Nałożyć wysokiej jakości system wiążący, zgodnie z instrukcją producenta.
3. Jeśli powierzchnia podkładu jest mała, na strzykawkę nałożyć końcówkę Black Micro;
w przypadku większych powierzchni użyć końcówki Black Mini.
4. Nałożyć preparat Ultra-Blend plus i przez 10 sekund utwardzać za pomocą lampy
polimeryzacyjnej VALO™ LED (w przypadku lamp o mocy <600mW/cm2 utwardzać
przez 20 sekund, a w przypadku lamp o mocy >600mW/cm2 - przez 10 sekund).
5. Jeśli ostateczny podkład ma być grubszy niż 2 mm, jako pierwszą warstwę nałożyć
najwyżej 0,5 mm i utwardzać 10 sekund. Według potrzeb dodawać dodatkowe
warstwy o grubości 2 mm.
Uwaga: preparat Ultra-Blend plus ulega utwardzeniu do głębokości 3 mm.
D. Przykrycie miazgi
UWAGA: Używać preparatu Ultra-Blend plus w przypadku odsłonięcia miazgi (różowej)
i małych struktur nieukrwionych. W przypadku odsłonięcia większych obszarów i/lub
odsłonięcia miazgiukrwionej należy rozpocząć leczenie endodontyczne.
1. Delikatnie pocierając za pomocą plastikowej końcówki Mini Dento-Infusor™ należy
nałożyć roztwór antybakteryjny Consepsis™ na 60 sekund. Nie płukać. Delikatnie
wysuszyć dmuchawką.
2. Preparat Ultra-Blend plus nałożyć za pomocą końcówki Black Micro na suchą zębinę
na granicy odsłonięcia, a następnie na odsłonięciu lub na cienkiej odsłoniętej
warstwie różowej zębiny.
UWAGA: należy ograniczyć pokrywanie zębiny do minimum w celu zmaksymalizowania
powierzchni dostępnej dla technik wiązania zębiny.
3. Przez 10 sekund utwardzać za pomocą lampy polimeryzacyjnej VALO™ LED (wprzypadku lamp o mocy <600 mW/cm2 utwardzać przez 20 sekund, a w przypadku
lamp o mocy >600 mW/cm2 - przez 10 sekund).
4. Nałożyć wysokiej jakości system wiążący, zgodnie z instrukcją producenta, i wypełnić
wybranym produktem.
E. Opaker
1. Zszorstkować metal za pomocą wiertła lub mikroabrazji.
2. Spłukać odsłonięty, przeznaczony do zamaskowania metal wodą i osuszyć
dmuchawką.
3. Nałożyć wysokiej jakości system wiążący, zgodnie z instrukcją producenta, iutwardzić.
4. Preparat Ultra-Blend plus nakładać na metal stopniowo w warstwach nie grubszych
niż 2 mm. Przez 10 sekund utwardzać za pomocą lampy polimeryzacyjnej VALO™
LED (w przypadku lamp o mocy <600 mW/cm2 utwardzać przez 20 sekund, a w
przypadku lamp o mocy >600 mW/cm2 - przez 10 sekund).
5. Uzupełnić kompozytem lub wybranym produktem.
Środki ostrożności:OSTRZEŻENIE: po kilku godzinach substancja może zestalić się w końcówce. Aby uniknąć
zatkania, należy zdjąć końcówkę i ponownie zamknąć osłonką łącznika Luer Lock pomiędzy
kolejnymi zastosowaniami. Należy zawsze sprawdzić wypływ materiału przed użyciem
w jamie ustnej. W przypadku wystąpienia oporu należy wymienić końcówkę i ponownie
skontrolować wypływ. Używać tylko zalecanych końcówek.
1. Przed użyciem preparatu Ultra-Blend plus należy dokładnie przeczytać i zrozumieć
wszystkie substancje i karty charakterystyki substancji. Stosować wyłącznie zgodnie
z zaleceniami.
2. W trakcie utwardzania preparatu Ultra-Blend plus lekarz i pacjent powinni nosić
okulary chroniące przed światłem UV.
3. W przypadku codziennego użytkowania przechowywać w temperaturze pokojowej.
Zapas schłodzić.
4. Aby zapobiec przedwczesnej polimeryzacji należy skierować górną lampę
stomatologiczną w innym kierunku.
5. Żywice mogą uczulać. Unikać wielokrotnego kontaktu nieutwardzonej żywicy
stomatologicznej ze skórą. Nie używać u pacjentów o stwierdzonej wrażliwości na
akrylany i inne żywice. W przypadku reakcji alergicznej, zapalenia skóry lub wystąpienia
wysypki należy skonsultować się z lekarzem.
6. Po użyciu końcówki i puste strzykawki odpowiednio zutylizować.
7. Przechowywać w miejscu niedostępnym dla dzieci.
8. Nie używać ponownie końcówek, aby uniknąć przeniesienia zakażenia.
9. Nie stosować po upływie terminu przydatności podanego na opakowaniu.
10. Aby uniknąć przeniesienia zakażenia między kolejnymi zastosowaniami należy
zdezynfekować i wytrzeć strzykawkę, stosując środek dezynfekujący o poziomieśrednim lub zastosować jednorazową osłonkę strzykawki. Jeśli używana jest osłonka
strzykawki, należy zdjąć końcówkę, zamknąć strzykawkę i wyrzucić końcówkę orazosłonkę strzykawki.
11. Preparat Ultra-Blend plus przechowywać z dala od źródeł ciepła/bezpośredniego
nasłonecznienia.

8 0 0 . 5 5 2 . 5 5 1 2牙本质垫底材料

RU© 2021 Ultradent Products, Inc. All Rights Reserved.

Ultra-Blend™ Plus

产品描述Ultra-Blend™ plus牙本质垫底材料是一种光固化X射线阻射材料，含有以二脲烷二甲基丙烯酸酯(UDMA)为基础材料的含氢氧化钙和羟基磷灰石成份的牙本质粘附垫底材料.牙本质垫底材料具有较高的填充量，可最大程度地减少收缩，并具有较高的粘度。
该产品可以用高速或低速手机轻松地勾勒出轮廓，有牙本质色和不透明色两种颜色选择。
结构组成该产品包括牙本质色和不透明白色2种颜色；由二脲烷二甲基丙烯酸酯(>59%)，二甲基丙烯酸三甘醇酯(>4%)，甘油二甲基丙烯酸磷酸盐(>5%)，樟脑醌(<0.2%)，甲基丙烯酸二甲氨乙酯(<0.5%)，硫酸钡(<10%)，氢氧化钙(<5%)，磷
酸钙(<15%)，二氧化钛(<1%)，和颜料(<0.08%)。
适用范围
该产品在放置复合树脂或银汞合金之前用作窝洞垫底和/或基底材料；无充血
性出血的牙髓附近或小面积露髓时用作牙髓封闭；冠修复时暴露金属的遮色。
使用说明
A．一般说明对于更大的预备区域，请选用Black
1. 将Black Micro™ 输送头与输送器拧紧。
Mini™ 输送头。
2. 根据厂商说明在预备区域使用合格的粘接材料。
3. 将0.5-2mm厚的Ultra-Blend plus涂抹在所需区域。
注意：当涂抹区域不易接触时，挤出少量产品在针输送头尖处形成药滴。
用输送头尖将药滴涂在指定部位，以防止过量涂抹。
4. 用VALO™ LED固化灯标准功率模式固化10秒（固化光输出功率<600mW/cm2
时固化20秒，固化光输出功率>600 mW /cm2固化10秒）。
注意：为了形成良好的基底层，建议首层涂抹厚度为0.5mm并光固化。
此后每层2mm涂抹并逐层固化。
5. Ultra-Blend plus可以用牙本质粘接剂粘附预备表面，也可以粘附于干燥的牙本质。
B. 牙本质垫底
1. 用水冲洗裸露的牙本质，而后吹干。
2. 根据厂商说明涂抹合格的粘接材料。
3. 用Black Micro输送头精准涂抹Ultra-Blend plus。
C. 保护基质
1.用水冲洗裸露的牙本质，而后吹干。
2. 根据厂商说明涂抹合格的粘接材料。
3. 基底区域较小的话使用Black Micro输送头；较大区域使用Black Mini输送头。
4. 涂抹Ultra-Blend plus后用VALO™ LED固化灯标准功率固化10秒（固化光输
出功率<600mW/cm2时固化20秒，固化光输出功率>600 mW /cm2固化10秒）。
5. 若最终基底厚于2mm，首层厚度0.5mm（或更小）并光固化10秒。
视需要继续添加，每层厚度2mm。
6. 注意：Ultra-Blend plus光固化深度为3mm。
D. 盖髓
注意：近髓（粉红色）及小型器械非充血性接触使用Ultra-Blend plus。
若大面积
接触和/或牙髓充血，则应进行根管治疗。
切
1. 用Mini Dento-Infusor™ 输送头擦洗式涂抹Consepsis™ 抗菌溶液60秒。
勿冲洗。
轻轻吹干。
2. 用Black Micro输送头在裸露面干燥的牙本质边缘处涂抹Ultra-Blend plus，而后在裸露面或近裸露面薄薄的粉红色牙本质部位涂抹。
注意：尽量减少牙本质覆盖，让牙本质粘接操作区域尽可能扩大。
3. 用VALO™ LED固化灯标准功率固化10秒（固化光输出功率<600mW/cm2时固
化20秒，固化光输出功率>600 mW /cm2固化10秒）。
4. 根据厂商说明涂抹合格的粘接材料，并使用所选产品修复。
E. 射线阻射1. 用牙钻或微研磨工具将打磨金属表面。
2. 用水冲洗不透射线的裸露金属面而后烘干。
3. 根据厂商说明涂抹合格的粘接材料后光固化。
4. 在金属表面逐层涂抹Ultra-Blend plus，每层不超过2mm厚。
用VALO™ LED固化灯标准功率固化10秒（固化光输出功率<600mW/cm2时固化20秒，固化光输
出功率>600 mW /cm2固化10秒）。
5. 用所选复合材料或产品进行修复操作。
注意事项警吿：几小时后化学品可能会在输送头处凝固。
为避免堵塞，每次使用后应换下输送头并拧上鲁尔锁帽。
在口腔内涂抹前始终检査出液是否顺畅。
如果遇到阻力，更换输送头并重新检査。
仅使用推荐输送头。
1. 使用Ultra-Blend plus之前仔细阅读理解所有说明和化学品安全数据表。
请严格按照说明使用。
有关产品的安全性数据表，可从官网查阅www.
ultradent.com
2. 固化Ultra-Blend plus时医生和患者均须佩戴紫外线护目镜。
3. 日常使用中须室温保存。
大批保存时需要冷藏。
4. 将头顶灯光照向别处，以防过早聚合。
5. 使用后的输送头和空注射器请按当地法规要求丢弃处理。
6. 切勿允许儿童接触。
7. 切勿重复使用输送头，以防交叉感染。
8. 切勿在容器标注的到期日后使用。
9. 为避免交叉感染，不同患者间使用时，使用Ultradent注射器帽和/或用中效
消毒剂擦拭注射器消毒。
若使用注射器帽，请取下输送头，重新拧上注射器帽，
并丢弃输送头和注射器帽。
10. 避免Ultra-Blend plus受热和阳光直射。
11. 仅供专业人士使用。
禁忌症
树脂材料可能导致过敏。
避免未固化牙科树脂与皮肤反复接触。
切勿用于
已知对丙烯酸酯或其他树脂敏感的患者。
若出现过敏反应、皮炎或皮疹，请
咨询医生。
储存条件
请参见产品标签。
生产日期和使用期限请参见产品标签。
配件清单含输送头（Black Micro tip 或Black Mini tip）
，输送头为一次性使用。
该配件的备案凭证号为国械备20181168号。
说明书修订日期
2021年05月17日
注册人和生产企业信息
Ultradent Products, Inc.
地址：505 West 10200 South South Jordan, Utah 84095, USA
电话：800.522.5512
代理人和售后服务单位
名称：广州市皓齿登医疗器械有限公司
地址：广州市海珠区工业大道北路67号34号楼首层自编102房
电话：020-84156505
产品名称：牙本质垫底材料
规格型号：色号：牙本质色、不透明色 规格：1.2ml/支
医疗器械注册证编号/产品技术要求编号：国械注进20163175083

Ultra-Blend™ PlusПрокладочный лайнерный и базовый материал на основе гидроксида кальция

Описание
Ultra-Blend™ plus представляет собой светоотверждаемый, рентгеноконтрастный, лайнерныйи азовый прокладочный материал, обладающий адгезией к дентину и содержащий
гидроксид кальция и гидроксиапатит кальция на основе уретандиметакрилата (УДМА).
Благодаря высокой степени наполнения материал Ultra-Blend plus обладает минимальной
усадкой, а его высокая вязкость облегчает нанесение. Данному материалу можно легко
придать форму с помощью высоко- или низкоскоростного наконечника бормашины.
Материал выпускается оттенка дентина и матово-белого оттенка.
Показания
Материал UltraBlend plus является самоклеящимся и используется в качестве прокладки подпломбу и (или) базового материала перед нанесением восстановительных материалов.
Указания по применению
A. Общие указания
1. Надежно закрепите насадку Black Micro™ на шприце. Для более крупных
препарируемых поверхностей используйте насадку Black
2. Mini™.
3. Подготовьте область с помощью качественной бондинговой системы в соответствиис инструкциями производителя.Нанесите слой материала Ultra-Blend plus толщиной
0,5–2 мм на требуемую область.
ПРИМЕЧАНИЕ. При нанесении на труднодоступную область выдавите небольшое
количество материала, чтобы на насадкеобразовался шарик. С помощью насадки
нанесите шарик на требуемую область во избежание выдавливания излишнего
количества материала.
4. Выполните отверждение светом в течение 10 секунд при стандартной мощности спомощью светодиодной полимеризационной лампы VALO™ (для ламп мощностью
<600 мВт/см2 время отверждения составляет 20 секунд, для ламп мощностью >600
мВт/см2 — 10 секунд).
ПРИМЕЧАНИЕ. Для создания базы сначала нанесите слой толщиной 0,5 мм ивыполните отверждение светом. Толщина последующих слоев должна составлять 2 мм сотверждением после нанесения каждого слоя.
5. Материал Ultra-Blend plus будет связываться с поверхностями, подготовленными спомощью бондов, обеспечивающих сцепление с дентином, и с сухим дентином.
B. Лайнерный материал для связывания с дентином
1. Промойте открытый дентин водой и высушите воздухом.
2. Нанесите качественную бондинговую систему в соответствии с инструкциями
производителя.
3. Нанесите материал Ultra-Blend plus точно и в желаемом объеме с помощью насадки
Black Micro.
4. Выполните отверждение светом в течение 10 секунд при стандартной мощности спомощью светодиодной полимеризационной лампы VALO™ (для ламп мощностью
<600 мВт/см2 время отверждения составляет 20 секунд, для ламп мощностью >600
мВт/см2 — 10 секунд).
C. Защитный базовый материал
1. Промойте открытый дентин водой и высушите воздухом.
2. Нанесите качественную бондинговую систему в соответствии с инструкциями
производителя.
3. Используйте насадку на шприц Black Micro, если наносить нужно на небольшой
участок, или Black Mini, если участок побольше.
4. Нанесите материал Ultra-Blend plus и выполните отверждение светом в течение 10секунд при стандартной мощности с помощью светодиодной полимеризационной
лампы VALO™ (для ламп мощностью <600 мВт/см2 время отверждения составляет 20секунд, для ламп мощностью >600 мВт/см2 — 10 секунд).
5. Если окончательная базовая прокладка должна быть толще 2 мм, нанесите первый
слой толщиной 0,5 мм (или менее) и выполните отверждение светом в течение 10секунд. По необходимости добавляйте дополнительные слои толщиной 2 мм.
Примечание. Отверждение светом для материала Ultra-Blend plus может выполнятьсядо глубины 3 мм.
D. Пульпарная прокладка
ПРИМЕЧАНИЕ. Используйте материал Ultra-Blend plus при поражении дентина внепосредственной близости к пульпе (розового цвета) и небольших механических
негиперемированных обнажениях пульпы. При более крупных обнажениях и (или)гиперемии пульпы следует начать эндодонтическое лечение.
1. Нанесите антибактериальный раствор Consepsis™ с помощью пластиковой насадки
Mini Dento-Infusor™ на 60 секунд аккуратными чистящими движениями. Непромывайте. Осторожно высушите воздухом.
2. Нанесите материал Ultra-Blend plus с помощью насадки Black Micro на сухой дентинпо границе с обнаженной пульпой, затем на саму обнаженную пульпу или на тонкий
розовый дентин в непосредственной близости от пульпы.
ПРИМЕЧАНИЕ. Старайтесь покрыть минимальную поверхность дентина, чтобы увеличить
доступную область для бондинга.
3. Выполните отверждение светом в течение 10 секунд при стандартной мощности спомощью светодиодной полимеризационной лампы VALO™ (для ламп мощностью
<600 мВт/см2 время отверждения составляет 20 секунд, для ламп мощностью >600
мВт/см2 — 10 секунд).
4. Нанесите качественную бондинговую систему в соответствии с инструкциями
производителя и выполните восстановление с помощью выбранного средства.
E. Опаковый материал
1. Придайте металлу шероховатость с помощью бора или процедуры микроабразии.
2. Ополосните водой и высушите воздухом обнаженный металл, подлежащий закрытию.
3. Нанесите качественную бондинговую систему в соответствии с инструкциями
производителя и выполните отверждение светом.
4. Поэтапно нанесите материал Ultra-Blend plus на металл, соблюдая толщину слоя неболее 2 мм. Выполните отверждение светом в течение 10 секунд при стандартной
мощности с помощью светодиодной полимеризационной лампы VALO™ (для лампмощностью <600 мВт/см2 время отверждения составляет 20 секунд, для лампмощностью >600 мВт/см2 — 10 секунд).
5. Выполните восстановление с помощью выбранного композита или средства.
Меры предосторожности
ПРЕДУПРЕЖДЕНИЕ. Через несколько часов работы на наконечнике могут скапливаться
химические вещества. Во избежание загрязнения снимайте насадку и надевайте
люэровские колпачки на шприц на время между использованием. Перед применением
средства в ротовой полости проверьте, насколько хорошо оно выдавливается. Еслисредство не проходит, замените насадку и повторите проверку. Используйте только
рекомендованные насадки.
1. Внимательно прочтите все указания и данные о безопасности материалов перед
использованием материала Ultra-Blend plus. Строго следуйте указаниям.
2. Во время отверждения материала Ultra-Blend plus медработник и пациент должны
использовать средства защиты глаз от УФ-излучения.
3. Количество продукта, необходимое для использования в течение дня, храните прикомнатной температуре. Основной запас следует хранить в холодильнике.
4. Перенаправьте верхний свет, чтобы предотвратить преждевременную полимеризацию.
5. Композиты могут повышать чувствительность. Избегайте частых контактов
неотвержденных стоматологических композитов с кожей. Не используйте препаратыдля лечения пациентов с повышенной чувствительностью к акрилатам и другим
композитам. При появлении аллергической реакции, дерматита или сыпи обратитеськ врачу.
6. Надлежащим образом утилизируйте использованные насадки и пустые шприцы.
7. Храните в недоступном для детей месте.
8. Во избежание перекрестного заражения не используйте насадки повторно.
9. Не используйте после истечения срока годности, указанного на упаковке.
10. Во избежание перекрестного заражения дезинфицируйте и протирайте шприц между
применениями дезинфицирующим средством средней активности и (или) используйте
одноразовые колпачки для шприца. При использовании колпачка для шприца снимите
насадку, наденьте на шприц новый колпачок и выбросьте насадку и старый колпачок.
11. Храните материал Ultra-Blend plus вдали от источников тепла и прямых солнечных лучей.

Manufactured by:
Ultradent Products, Inc.
505 West Ultradent Drive (10200 South)
South Jordan, Utah 84095 USA
Manufactured in the USA from globally sourced materials
Ultradent Products GmbHAm Westhover Berg 30
51149 Cologne GermanyKOELΠεριγραφή:Το Ultra-Blend™ plus είναι ένα φωτοπολυμερισμένο, ακτινοσκιερό επενδυτικό υλικό βάσηςπου κολλάει στην οδοντίνη και περιέχει υδροξείδιο του ασβεστίου και υδροξυαπατίτη
ασβεστίου σε βάση διμεθυλακρυλικής ουρεθάνης (UDMA). Το Ultra-Blend plus έχει έναφορτίο υψηλής πλήρωσης προς αποφυγή συρρίκνωσης και υψηλότερο ιξώδες για ναδιευκολύνει την τοποθέτηση. Μπορεί να διαμορφωθεί εύκολα με χειρολαβή υψηλής ήχαμηλής ταχύτητας και διατίθεται σε αποχρώσεις οδοντίνης και αδιαφανούς άσπρου
χρώματος.
Ενδείξεις:Το Ultra-Blend plus κολλάει από μόνο του και χρησιμοποιείται ως επενδυτικό υλικό
κοιλότητας ή/και υλικό βάσης πριν από την τοποθέτηση υλικών αποκατάστασης.
Οδηγίες χρήσης:A. Γενικά
1. Συνδέστε το άκρο Βlack Micro™ ασφαλώς επάνω στη σύριγγα. Για μεγαλύτερες
προετοιμασίες, χρησιμοποιήστε το άκρο Black Mini™.
2. Προετοιμάστε την περιοχή με ένα ποιοτικό σύστημα συγκόλλησης σύμφωνα με τιςοδηγίες του κατασκευαστή.
3. Απλώστε μια στρώση 0,5-2 χιλ. από Ultra-Blend plus στην περιοχή που θέλετε.
ΣΗΜΕΙΩΣΗ: Όταν το τοποθετείτε σε μια περιοχή στην οποία είναι δύσκολη η πρόσβαση,βγάλτε μια μικρή ποσότητα για να σχηματιστεί ένα μπαλάκι στο άκρο.
4. Χρησιμοποιήστε το άκρο ως απλικατέρ μπάλας για να αποτρέψετε την υπερβολική
παροχή. Εκτελείτε φωτοπολυμερισμό με το φως πολυμερισμού VALO™ LED υπόΤυπική Ισχύ επί 10 δευτερόλεπτα (για φώτα με παραγωγή <600mW/cm2 εκτελείτε
πολυμερισμό επί 20 δευτερόλεπτα ή για φώτα με παραγωγή >600mw/cm2 εκτελείτε
πολυμερισμό επί 10 δευτερόλεπτα).
ΣΗΜΕΙΩΣΗ: Για ενίσχυση της βάσης, ξεκινήστε με μια στρώση 0,5 χιλ. και εκτελέστε
φωτοπολυμερισμό. Συνεχίστε με διαστήματα των 2 χιλ. εκτελώντας πολυμερισμό
ανάμεσα σε κάθε διάστημα.
5. Το Ultra-Blend plus θα συνδεθεί με τις επιφάνειες που έχουν προετοιμαστέι μεπαράγοντες συγκόλλησης οδοντίνης και με στεγνή οδοντίνη.
B. Επενδυτικό υλικό οδοντίνης
1. Ξεπλένετε την εκτεθειμένη οδοντίνη με νερό και τη στεγνώνετε με αέρα.
2. Απλώστε ένα ποιοτικό σύστημα συγκόλλησης σύμφωνα με τις οδηγίες τουκατασκευαστή.
3. Τοποθετήστε το Ultra-Blend plus επακριβώς και με το σωστό όγκο που θέλετε
χρησιμοποιώντας ένα άκρο Black Micro.
4. Εκτελείτε φωτοπολυμερισμό με Φως Πολυμερισμού VALO™ LED υπό Τυπική Ισχύ επί
10 δευτερόλεπτα (για φώτα με παραγωγή <600mW/cm2 εκτελείτε πολυμερισμό επί
20 δευτερόλεπτα ή για φώτα με παραγωγή >600mw/cm2 εκτελείτε πολυμερισμό επί
10 δευτερόλεπτα).
C. Προστατευτική βάση1. Ξεπλένετε την εκτεθειμένη οδοντίνη με νερό και τη στεγνώνετε με αέρα.
2. Απλώστε ένα ποιοτικό σύστημα συγκόλλησης σύμφωνα με τις οδηγίες τουκατασκευαστή.
3. Συνδέστε ένα άκρο Black Micro στη σύριγγα εάν η περιοχή της βάσης είναι μικρή.
Χρησιμοποιήστε ένα άκρο Black Micro για μεγαλύτερες περιοχές.
4. Απλώστε το Ultra-Blend plus και εκτελέστε φωτοπολυμερισμό με το ΦωςΠολυμερισμού VALO™ LED υπό Τυπική Ισχύ επί 10 δευτερόλεπτα (για φώτα με
παραγωγή <600mW/cm2 εκτελείτε πολυμερισμό επί 20 δευτερόλεπτα ή για φώτα με
παραγωγή >600mw/cm2 εκτελείτε πολυμερισμό επί 10 δευτερόλεπτα).
5. Εάν η τελική βάση πρέπει να έχει πάχος περισσότερο από 2 χιλ., απλώστε 0,5 χιλ. (ήλιγότερο) ως πρώτη στρώση και εκτελέστε φωτοπολυμερισμό επί 10 δευτερόλεπτα.
Προσθέστεεπιπρόσθετες στρώσεις πάχους 2 χιλ. όπως χρειάζεται.
Σημείωση: Το Ultra-Blend plus φωτοπολυμερίζει σε βάθος των 3 χιλ.
D. Κάλυψη πολφού
ΣΗΜΕΙΩΣΗ: Χρησιμοποιήστε το Ultra-Blend plus για εκτεθειμένο ιστό κοντά στον πολφό(ροζ) και μικρή μηχανική έκθεση χωρίς υπεραιμία. Εάν υπάρχει μεγαλύτερη έκθεση ή/καιπολφός υπεραιμίας, πρέπει να ξεκινήσετε ενδοδοντική θεραπεία.
1. Απλώστε αντιβακτηριδιακό διάλυμα Consepsis™ με πλαστικό άκρο Mini Dento-Infusor™
επί 60 δευτερόλεπτα με απαλό βούρτσισμα. Μην το ξεπλένετε. Στεγνώστε το απαλά.
2. Απλώστε το Ultra-Blend plus με το άκρο Black Micro στη στεγνή οδοντίνη στοπεριθώριο της έκθεσης και μετά επάνω στην έκθεση ή επάνω από λεπτή ροζ οδοντίνητης κοντινής. έκθεσης.
ΣΗΜΕΙΩΣΗ: Ελαχιστοποιείτε την κάλυψη της οδοντίνης για να μεγιστοποιήσετε τηνπεριοχή που είναι διαθέσιμη για τεχνικές συγκόλλησης οδοντίνης.
3. Εκτελείτε φωτοπολυμερισμό με Φως Πολυμερισμού VALO™ LED υπό Τυπική Ισχύ επί
10 δευτερόλεπτα (για φώτα με παραγωγή <600mW/cm2 εκτελείτε πολυμερισμό επί
20 δευτερόλεπτα ή για φώτα με παραγωγή >600mw/cm2 εκτελείτε πολυμερισμό επί
10 δευτερόλεπτα).
4. Απλώστε ένα ποιοτικό σύστημα συγκόλλησης σύμφωνα με τις οδηγίες τουκατασκευαστή και εκτελείτε την αποκατάσταση με το προϊόν που προτιμάτε.
E. Σύστημα αδιαφάνειας
1. Τραχύνετε το μέταλλο με φρέζα ή με μικροαδροποίηση.
2. Ξεπλένετε το εκτεθειμένο μέταλλο το οποίο πρόκειται να γίνει αδιαφανές με νερό καιτο στεγνώνετε με αέρα.
3. Απλώστε ένα ποιοτικό σύστημα σύμφωνα με τις οδηγίες του κατασκευαστή καιεκτελέστε φωτοπολυμερισμό.
4. Απλώστε το Ultra-Blend plus επάνω από το μέταλλο σε διαστήματα αλλά προσέξτε ναμην έχει καμία στρώση περισσότερο πάχος από 2 χιλ. Εκτελείτε φωτοπολυμερισμό μετο Φως Πολυμερισμού VALO™ LED υπό Τυπική Ισχύ επί 10 δευτερόλεπτα (για φώτα με
παραγωγή <600mW/cm2 εκτελείτε πολυμερισμό επί 20 δευτερόλεπτα ή για φώτα με
παραγωγή >600mw/ cm2 εκτελείτε πολυμερισμό επί 10 seconds).
5. Εκτελείτε την αποκατάσταση με το σύνθετο ή το προϊόν που προτιμάτε.
Προφυλάξεις:ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Η χημική ουσία ενδέχεται να σταθεροποιηθεί εντός του άκρου μετά απόμερικές ώρες. Για να αποφύγετε το φράξιμο, αφαιρέστε το άκρο καιεπανατοποθετήστε τα πώματα Luer Lock μεταξύ των χρήσεων. Επαληθεύστε πάντοτε τηροή του υλικού προτού το χρησιμοποιήσετε ενδοστοματικά. Εάν αισθανθείτε αντίσταση,αντικαταστήστε το άκρο και ελέγξτε το ξανά. Χρησιμοποιήστε μόνο συνιστώμενα άκρα.
1. Διαβάστε προσεκτικά και κατανοήστε όλες τις οδηγίες και το Φύλλο Δεδομένων
Ασφαλείας Υλικών (SDS) προτού χρησιμοποιήσετε το Ultra-Blend plus. Χρησιμοποιήστετο μόνο σύμφωνα με τις οδηγίες χρήσης.
2. Ο ιατρός και ο ασθενής πρέπει να φοράνε προστατευτικά γυαλιά έναντι των υπεριωδών
ακτίνων όταν εκτελείται πολυμερισμός με Ultra-Blend plus.
3. Για καθημερινή χρήση, αποθηκεύστε το σε θερμοκρασία δωματίου.
4. Τοποθετήστε το χύμα υλικό στο ψυγείο.
5. Επαναπροσανατολίστε το υπερυψωμένο φως για να αποτρέψετε τον πρώιμο
πολυμερισμό. Οι ρητίνες ενδέχεται να προκαλέσουν ευαισθητοποίηση. Αποφύγετε τηνεπαναλαμβανόμενη επαφή από απολυμέριστες οδοντιατρικές ρητίνες με το δέρμα. Μηντο χρησιμοποιείτε σε ασθενείς με γνωστή ευαισθησία σε ακρυλικές ή άλλες ρητίνες. Εάνεμφανιστεί αλλεργική αντίδραση, δερματίτιδα ή εξάνθημα, συμβουλευτείτε ιατρό.
6. Πετάξτε τα χρησιμοποιημένα άκρα και τις άδειες σύριγγες με τον κατάλληλο τρόπο.
7. Φυλάξτε το μακριά από τα παιδιά.
8. Προς αποφυγή διασταυρούμενης μόλυνσης, μην επαναχρησιμοποιείτε τα άκρα.
9. Μην το χρησιμοποιείτε μετά την ημερομηνία λήξης που αναγράφεται στα δοχεία.
10. Προς αποφυγή διασταυρούμενης μόλυνσης, απολυμαίνετε και σκουπίζετε τη σύριγγα
με ένα απολυμαντικό μεσαίου επιπέδου μεταξύ χρήσεων ή/και χρησιμοποιήστε ένααναλώσιμο κάλυμμα σύριγγας. Εάν χρησιμοποιηθεί κάλυμμα στη σύριγγα, αφαιρέστε τοάκρο, επανατοποθετήστε το πώμα, και πετάξτε το άκρο και το κάλυμμα της σύριγγας.
11. Φυλάξτε το Ultra-Blend plus μακριά από θερμότητα/ άμεσο φως ηλίου.

Ultra-Blend™ Plus

Ultra-Blend™ Plus수산화칼슘(소석회) 상아질 라이너 및 프로텍티브 베이스설명:
Ultra-Blend™ 플러스는 취화 칼슘 우레탄(UDMA)계에 수산화칼슘(소석회)및 칼슘 수산화인회석을 함유하는 광중합의 방사선 불투과성 상아질 접착형라이너 및 베이스 자재입니다. Ultra-Blend plus는 수축을 최소화하기 위한많은 양의 충전재 로딩과 손쉽게 설치하기 위한 높은 점성을 가집니다. 고속또는 저속 핸드피스로 윤곽처리가 가능하며, 상아질과 불투명성 백색 색조로이용이 가능합니다.
지시사항:
Ultra Blend plus는 자가접착형이며 우식 라이너 및/또는 수복물 자재를설치하기 전에 모재로서 사용됩니다.
사용지침:
A. 일반사항
1. Black Micro™ 팁을 주사기에 확실하게 부착합니다. 대형 조제물의 경우,Black Mini™ 팁을 사용합니다.
2. 제조업체의 안내사항에 따라 해당 부위를 고품질 접착제로 조제합니다.
3. 원하는 부위의 표면에 Ultra-Blend 플러스를 0.5-2mm 두께의 층으로도포합니다.
주: 소량을 종이 패드 위에 압착하여 사용하기 전에 팁을 통해 흐름상태가
균일한지 여부를 확인합니다. 팁을 구형 도포용 도구로 사용하여 과다 공급(디스펜싱)을 방지합니다.
4. 표준 성능으로는 10초 간 VALO™ LED 광중합기를 사용하여 광중합을실시합니다(광선 출력 <600mW/cm2의 경우, 20초 간 또는 광선 출력>600mw/cm2의 경우 10초 간 중합 실시).
주: 베이스를 구축할 경우, 0.5mm의 층으로부터 시작하여 광중합을실시합니다. 2mm 점증작업을 실시하고는 각 점증작업 간에 중합을실시합니다.
5. Ultra-Blend 플러스는 상아질 접착제로 조제한 표면과 마른 상태의상아질에 접착하게 됩니다.
B. 상아질 라이너
1. 물로 노출된 상아질을 헹구고 공기건조를 실시합니다.
2. 제조업체의 안내사항에 따라 고품질 접착 시스템을 도포합니다.
3. Black Micro 팁으로 Ultra-Blend 플러스를 정밀하며 원하는 양을정확하게 사용합니다.
4. 표준 성능으로는 10초 간 VALO™ LED 광중합기를 사용하여 광중합을실시합니다(광선 출력 <600mW/cm2의 경우, 20초 간 또는 광선 출력>600mw/cm2의 경우 10초 간 중합 실시).
C. 포로텍티브 베이스
1. 노출된 상아질을 물로 헹구고 공기건조를 실시합니다.
2. 제조업체의 안내사항에 따라 고품질 접착 시스템을 도포합니다.
3. 베이스 부위가 좁을 경우, Black Micro 팁을 주사기에 부착하고, 넓은부위에는 Black Mini 팁을 사용합니다.
4. Ultra-Blend 플러스를 도포하고 표준 성능으로는 10초 간 VALO™ LED광중합기를 사용하여 광중합을 실시합니다(광선 출력 <600mW/cm2의
경우, 20초 간 또는 광선 출력 >600mw/cm2의 경우 10초 간 중합 실시).
5. 최종 베이스가 2mm 두께 이상일 경우, 첫 번재 층으로서 0.5mm (또는이하) 두께로 도포하고 10초 간 중합을 실시합니다필요에 따라 2mm
두께의 층을 추가합니다.
주: Ultra-Blend 플러스는 3mm의 깊이까지 광중합을 실시합니다.
D. 치수 캐핑주: 치수(분홍색) 근처에는 Ultra-Blend 플러스와 소량의 기계적 비
hyperemic 노출을 사용합니다. 노출부위가 크며 또는 충혈된 치수의 경우,치내요법(신경치료법)을 시작해야 합니다.
1. 플라스틱 Mini Dento-Infusor™ 팁으로 Consepsis™ 항균액을 60초동안 부드럽게 문지르듯 하면서 도포합니다. 헹구지 마십시오. 부드럽게공기로 말립니다.
2. Black Micro 팁을 사용해 Ultra-Blend 플러스를 건조시킨 상아질에노출부위 경계를 따라, 그 다음에는 노출부위 위나 인근 노출부위의 분홍빛상아질 위에 도포합니다.
주: 상아질 접착기법에 대하여 해당 부위를 최대한 접근할 수 있도록 상아질의덮는 작업을 위소화합니다.
3. 표준 성능으로는 10초 간 VALO™ LED 광중합기를 사용하여 광중합을실시합니다(광선 출력 <600mW/cm2의 경우, 20초 간 또는 광선 출력>600mw/cm2의 경우 10초 간 중합 실시).
4. 제조업체의 안내사항에 따라 고품질 접착 시스템을 도포하고 정선된제품을 사용해 수복합니다.
E. Opaquer(오페커)1. 버 또는 미세연마를 통해 금속부위를 거칠게 처리합니다.
2. 노출된 금속부위를 물로 헹구어 불투명하기 하고는 공기건조를
실시합니다.
3. 제조업체의 안내사항에 따라 고품질 접착 시스템을 도포한 다음, 광중합을실시합니다.
4. Ultra-Blend 플러스를 금속부위 위에 층이 2mm 를 넘지 않는 두께로점증식의 도포를 실시합니다. 표준 성능으로는 10초 간 VALO™ LED광중합기를 사용하여 광중합을 실시합니다(광선 출력 <600mW/cm2의
경우, 20초 간 또는 광선 출력 >600mw/cm2의 경우 10초 간 중합 실시).
5. 복합레진(컴포짓)이나 정선된 제품을 사용해 수복합니다.
주의사항:경고사항: 몇 시간이 지나면 화학물질이 팁에 고착될 수 있습니다. 막히지 않도록하기 위해서는 사용 간에 팁을 제거하고 루어 락 캡을 다시 씌웁니다. 구강내에사용하기 전에 자재의 흐름상태를 항상 확인합니다. 저항감이 느껴지면, 팁을교체한 후 다시 점검합니다. 권장하는 팁에 한해 사용합니다.
1. Ultra-Blend 플러스를 사용하기 전에 일체의 안내사항 및 SDS(
물질안전보건자료)를 주의하여 읽고 숙지합니다. 지시내용에 따라서만사용합니다.
2. Ultra-Blend 플러스를 중합시키는 동안, 임상의와 환자는 자외선용보안경을 착용해야 합니다.
3. 매일 사용하는 경우에는 실온에 보관하십시오. 대량 보관 시에는 냉장합니다.
4. 광중합이 덜 되지 않도록 오버헤드 중합기의 방향을 조정합니다.
5. 레진은 감작의 가능성이 있습니다. 중합되지 않은 치과용 레진이 피부에반복적으로 접촉하지 않도록 합니다. 아크릴산 또는 기타 레진에 과민증이밝혀진 환자에게는 사용하지 마십시오. 알러지 반응, 피부염 또는 발진이진행할 경우, 의사의 진찰을 받습니다.
6. 사용한 팁과 빈 주사기는 적절히 처리합니다.
7. 어린이의 손길이 닿지 않는 곳에 보관합니다.
8. 교차오염을 피하려면, 팁은 재사용하지 마십시오.
9. 용기에 표시된 유효기간 후에는 사용하지 마십시오.
10. 교차오염을 방지하기 위해서는, 사용 간에는 중간 정도의 소독제로 주사기를소독 및 닦아 주고/또는 1회용 주사기 커버를 사용합니다. 주사기 커버를사용하는 경우, 팁을 제거하고 캡을 다시 씌운 후 팁과 주사기 커버는폐기합니다.
11. Ultra-Blend 플러스는 열/햇빛이 직접 들지 않는 곳에 보관합니다.

Ultra-Blend™ PlusΕπενδυτικό υλικό οδοντίνης και προστατευτική βάση από υδροξείδιο του ασβεστίου

Calcium Hydroxide Dentin Liner and Protective BaseNO

HealthUltra-Blend™ Plus

Kalsiumhydroksid tannfôring og beskyttelsesbase

Beskrivelse:
Ultra-Blend™ plus er et lysherdende, røntgentett fôringsmiddel og basemateriale
med kalsiumhydroksid og kalsiumhydroksyapatitt i en uretandimetakrylat (UMDA)base. Ultra-Blend plus har høy fyllkapasitet for å minimiere innkrymping og en
høyere viskositet for å lette påføringen. Den kan lett formes med et høy- eller
lavhastighets håndstykke og er tilgjengelig i dentin og opak hvitnyanser.
Indikasjoner:
UltraBlend plus er selvheftende og brukes som en hulromsfôring og/eller etbasemateriale før påføring av restaureringsmaterialer.
Retningslinjer for bruk:
A. Generelt
1. Fest Black Micro™-spissen godt fast på sprøyten. For større preparasjoner,
bruk Black Mini™-spiss.
2. Klargjør området med et kvalitets bindingssystem i henhold til produsentens
anvisninger.
3. Før på et 0,5-2 mm lag Ultra-Blend plus på det ønskede området.
MERK: Ved påføring på områder som det er vanskelig å få tilgang til, trykk ut littmateriale slik at det danner en ball på spissen. Bruk spissen til å føre på en liten
mengde og unngå overdosering.
4. Lysherd med VALO™ LED-herdelys på Standard Power i 10 sekunder (for lys
med utgang <600 mW/cm2 herde i 20 sekunder eller for lys med utgang>600 mw/cm2 herde i 10 sekunder).
MERK: For baseoppbygging, start med et 0,5 mm tynt lag og lysherd. Følg opp
med 2 mm lag og herde mellom hvert lag.
5. Ultra-Blend plus vil binde til overflater klargjort med dentinbindemidlerog til tørt dentin.
B. Dentinfôring
1. Skyll eksponert dentin med vann og lufttørk.
2. Påfør et kvalitets bindingssystem i henhold til produsentens anvisninger.
3. Før på Ultra-Blend plus nøyaktig og i eksakt ønsket mengde med en Black
Micro-spiss.
4. Lysherd med VALO™ LED-herdelys på Standard Power i 10 sekunder (for lys
med utgang <600 mW/cm2 herde i 20 sekunder eller for lys med utgang>600 mw/cm2 herde i 10 sekunder).
C. Beskyttelsesbase
1. Skyll eksponert dentin med vann og lufttørk.
2. Påfør et kvalitets bindingssystem i henhold til produsentens anvisninger.
3. Bruk en Black Micro-spiss sprøyte dersom baseområdet er lite. Bruk en Black
Mini-spiss for større områder.
4. Påfør Ultra-Blend plus og lysherd med VALO™ LED-herdelys på Standard
Power i 10 sekunder (for lys med utgang <600 mW/cm2 herde i 20 sekundereller for lys med utgang >600 mw/cm2 herde i 10 sekunder).
5. Dersom den endelige basen er ment å være mer enn 2 mm tykk, påfør 0,5mm (eller mindre) for det første laget og lysherd i 10 sekunder. Legg til
ytterligere lag i 2 mm tykkelse etter behov.
Merk: Ultra-Blend plus kan lysherdes til en dybde på 3 mm.
D. D. Rotkanalfylling
MERK: Bruk Ultra-Blend plus nær tannkjøtt (rosa) og liten mekanisk ikkehypermisk eksponering. Hvis større eksponering og/eller hvis hypermisk
tannkjøtt, bør endodontisk behandling startes.
1. Påfør Consepsis™ antibakteriell oppløsning med plastisk Mini DentoInfusor™-spiss i 60 sekunder ved bruk av en lett skurende bevegelse. Ikke
skyll. Lufttørk forsiktig.
2. Påfør Ultra-Blend plus med Black Micro-spiss på det tørre dentinet ved
kanten for eksponeringsområdet og deretter over eksponeringsområdet
eller over det tynne, rosa dentinet ved næreksponering.
MERK: Minimer dentintildekking for å øke tilgjengelig område for
dentinbindingsteknikker.
3. Lysherd med VALO™ LED-herdelys på Standard Power i 10 sekunder (for lys
med utgang <600 mW/cm2 herde i 20 sekunder eller for lys med utgang>600 mw/cm2 herde i 10 sekunder).
4. Påfør et kvalitets bindingssystem i henhold til produsentens anvisninger og
restaurer med foretrukket produkt.
E. Avdekning
1. Bruk bor eller mikroabrasjon til å gjøre metallet ujevnt.
2. Skyll eksponert metall som skal avdekkes med vann og lufttørk.
3. Påfør et kvalitets bindingssystem i henhold til produsentens anvisninger
og lysherd.
4. Påfør Ultra-Blend plus over metallet i maksimalt 2 mm tykke lag. Lysherd
med VALO™ LED-herdelys på Standard Power i 10 sekunder (for lys med
utgang <600 mW/cm2 herde i 20 sekunder eller for lys med utgang >600
mw/cm2 herde i 10 sekunder).
5. Restaurer med kompositt eller annet ønsket produkt.
Forholdsregler:
ADVARSEL: Kjemikalier vil herdes i spissen etter noen timer. For å unngå
tilstopping, fjern spissen og lukk ved bruk av luerlåshetten mellom hver bruk.
Materialflyt må alltid bekreftes før intraoral bruk. I tilfelle motstand, skift ut
spissen og sjekk igjen. Bruk kun anbefalte spisser.
1. Forsikre deg om at du nøye har lest og forstått SDS før bruk av Ultra-Blend
plus. Bruk kun som anvist.
2. Kliniker og pasient må bruke UV-beskyttelsesbriller under herding av
Ultra-Blend plus.
3. For daglig bruk, oppbevar i romtemperatur. Lagerbeholdning må oppbevaresi kjøleskap.
4. Snu vekk arbeidslys for å hindre for tidlig polymerisering.
5. Resiner kan forårsake følsomhet. Unngå gjentatt hudkontakt med uherdet
dentalresin. Ikke bruk på pasienter med en kjent følsomhet til akrylater eller
andre resiner. I tilfelle allergisk reaksjon, utvikling av dermatitt eller utslett,kontakt lege.
6. Kast brukte spisser og tomme sprøyter forskriftsmessig.
7. Oppbevares utilgjengelig for barn.
8. For å unngå krysskontaminering, ikke bruk spissene mer enn én gang.
9. Ikke bruk etter utløpsdato som er merket på beholderne.
10. For å unngå smitteoverføring, desinfiser og tørk av sprøyten med
et mellomnivå desinfiseringsmiddel mellom bruk og/eller bruk etsprøyteovertrekk til engangsbruk. Dersom det brukes sprøyteovertrekk, fjern
spissen, sett hetten tilbake på plass, kast spissen og sprøyteovertrekket.
11. Hold Ultra-Blend plus vekke fra varme/direkte sollys.