3D Accuitomo
170

Thinking ahead. Focused on life.

3D Accuitomo 170

Unsurpassed image clarity
The 3D Accuitomo 170 offers unsurpassed image clarity. With 9 fields of view and multiple acquisition modes, the 3D
Accuitomo 170 can meet all of your diagnostic needs with unparalleled quality. Its super-fine minimal voxel size of just 80µm
allows diagnosing even the most subtle details of bone and dentition. The 3D Accuitomo 170 is highly recommended byleading dental radiologists for periodontology, oral surgery, endodontics, orthodontics, dental implants, for the maxillofacial
region and beyond.

ø170 × H 120 mm (250μm)

ø40 × H 40 mm (80μm)

Stunning clarity
The high resolution 80μm acquisitions provide images of stunning clarity, giving you diagnostic information like you’ve neverseen before. Take your treatment planning to the next level.

High-resolution 360° (80μm)
2Standard-resolution 360° (80μm)

High-speed 360° (80μm)

Adaptable Acquisition Modes

The 3D Accuitomo 170 is equipped with four imaging modes that allow flexibility when scanning patients with a variety ofdiagnostic needs and clinical indications. Choose High Resolution and High Fidelity modes for the best quality images, orHigh Speed for patients that have difficulty remaining still. The 3D Accuitomo 170 will adapt to suite your diagnostic andyour patients’ needs.

180°

High-Speed

Standard Mode
360º scan 17.5 sec, 180º scan: 9 sec
Standard mode offers images of exceptional clarity and issuitable for limited and wide views of temporal bone,paranasal, sinus, maxilla and mandible, individual teeth, etc.

5.4 sec
9 sec

Standard

10.5 sec

High-Speed

High Resolution Mode
18 sec At 1/4 the standard pixel size, high-resolution mode proStandard
duces the sharpest and clearest images the 3D Accuitomo 170※ High-Speed Mode is available for ø 40 x H40mm and ø 60 x H60mm fields of has to offer. Even in hi-res mode, 360 scans take only 30.8 sec,view only.
and 180 scans a mere 15.8 sec. Available for 40 x 40 mm
and 60 x 60 mm FOVs.
360°

9 different fields of view can be selected to meet a wide
variety of clinical needs. From the smallest, 40 x 40 to thelargest 170 x 120, there is a size that fits your needs while
always ensuring the lowest X-ray doses possible to thepatient. This flexibility allows the 3D Accuitomo 170 toprovide stunning images for Endodontics, Periodontics,Maxillofacial surgery and many more.

High Fidelity Mode
Slow and steady scans at 30.8 sec for 360° and 15.8 sec
for 180° makes for exceptionally clear images with minimalartifacts. Zoom reconstructions made from this acquisition
are exceptionally clear.
High Speed Mode360 scan: 10.5 sec, 180 scan: 5.4 sec utilize high speed
mode to reduce motion artifacts for patients that may not
be able to sit still. It is a good choice for children and forpatients concerned with higher X-ray dosage. Available for40 x 40mm and 60 x 60 mm FOVs.

ø170 mm

ø170 mm

ø100 mm
ø80 mm
ø60 mm
ø40 mm

H120 mm

ø100 mm

H100 mm
H80 mm
H60 mm
H50 mm
H40 mmø140 mm

ø140 mm

ø100 mm

ø80 mm
ø60 mm
ø40 mm3

Small Fields of View

Stay focused on your region of interest by selecting the best field of view for your indication.
Volume diameters as small as 40 mm or as large as 100 mm can be selected for the dentition.

ø100 mm
ø80 mm
ø60 mm
ø40 mm1. ø100 × H100 mm (250μm)

2. ø60 × H60 mm (125μm)
4

3. ø40 × H40 mm (80μm)

1. H100 mm
2. H80 mm
3. H60 mm
4. H40 mm

Large Fields of View

For larger maxillofacial scans, select a diameter of ø100 to ø170 to cover a wider rangeof maxillofacial surgeries.

ø170 mm
ø140 mm

5. H120 mm
6. H100 mm

ø100 mm

7. H50 mm

4. ø170 × H120 mm (250μm)

5. ø140 × H100 mm (250μm)

6. ø100 × H50 mm (250μm)
5Simple, Accurate Positioning

The three positioning laser beams and an LCD make patient
positioning easy.
The chinrest stabilizes the patient`s head to avoid movement. Scoutimages enable even more accurate positioning.
Easy as One, Two, Three.
First, the patient`s initial position is set and recorded using the threepositioning laser beams. Then, the region of interest is aligned in theLCD. The chair automatically moves into the optimal position. During
the X-ray exposure, the patient is stabilized by the chinrest and theheadrest.

X Cursor
Z Cursor

Y Cursor

2–direction scout
For even more accurate positioning, scout images can be utilized.
After positioning, two still X-ray images of coronal and sagittal views
can be taken to confirm that the position is accurate. If adjustment isnecessary, positioning can be changed by dragging the cursor on themonitor and moving it to the center of the region of interest.
The scout exposure (80kV and 2.0 mA) will increase the total X-ray
dosage of a Standard Mode CT exposure (90kV and 5.0 mA) by
about 2%.

3D-CT image
Region of interest isperfectly positioned.

6

Acquisition is only the beginning180 degree reconstruction to reduce
motion artifacts
Did the patient move during acquisition?
Do you have to scan them again? Not
any more! The 3D Accuitomo 170 allows
you to extract 1/2 of a 360° scan at any
point to remove that initial jolt or thatunintended swallow at the end of theacquisition.
(For 360 ° acquisitions only)

Remove unintended motion artifacts
180°Patient moved

360° Scan
Extract 180° motion-free data
360° scan exhibiting patient movement artifacts

180° of motion-free data extracted.

Zoom reconstruction
The Accuitomo series is equipped witha unique zoom reconstruction function
allowing you to zoom in andreconstruct a new volume from theoriginal scan, without the need foradditional acquisitions. The newvolume can be reconstructed with aresolution of up to 80μm improving
diagnostic accuracy with no additional
X-ray exposure to the patient.

Ø170 × H120 mm (250μm)

Ø40 × H40 mm (80μm)

7i-Dixel

Acquisition to diagnosis made simple
The i-Dixel imaging software offers awide variety of features to help you
quickly and easily createcomprehensive treatment plans andexplain those plans to your patients.
Mandibular canal marking, implantpresentation, multiplanarreconstruction are just a few of thefeatures that i-Dixel provides fordiagnoses. i-Dixel is also fully DICOM
compliant and provides quick and easy
integration with both practicemanagement software and advanced
treatment planning tools.

Volume Rendering
Volume rendering displays a solid 3D image showing the bone structure anddental arch. The volume rendered image is linked to the slices and moves in realtime whenever a slice is adjusted.

From multiplanar reconstructions to implant presentations
Implant Presentation
Select from a variety of popular implant manufacturers and place the implant
directly in the CT slice for presentation. This presentation can be easily
understood by patients and helps with case acceptance.
8i-Dixel WEB

Freedom from platform and simplicity of designi-Dixel WEB runs as a web service on an X-ray server PC included with your Morita X-ray system. It serves as a local andsecure web-based dental image processing service that you can access throughout your practice on a wide range of devices.

No software installation neededWith the latest advancements in
web technology, i-Dixel WEB gives
you the freedom to view your images
wherever you want and however youwant. Gone are the days of complicated chairside PC setups, and limited
choices of hardware. Mac OS X andeven iPads can be used to view andedit data from a Morita X-ray system.

3D image processing
Even though i-Dixel WEB is a fully
web-based system, the features
you’ve come to expect are not compromised. View Morita’s signaturehigh quality images on the device ofyour choice!

Mac, iPad, Safari, OS X and Mac OS are trademarks of Apple Inc., registered in the U.S. and othercountries. Windows, Windows 8, Windows 10 and Internet Explorer are registered trademarks ofMicrosoft Corporation in the United States and other countries.

9Implantology
Case 1: Female patient referred for
3-dimensional analysis of an esthetic
complication in the left maxillary incisor
region (a).
The clinical status exhibits a mucosal
recession as well as a flattening anddiscoloration of the facial mucosa at theimplant crown. The patient complained
about recurrent peri-implant infections.
* : Nasal palate tube

Image 1a: Clinical aspect at the initial examination

Image 1c: Axial CBCT slice

Image 1b: Coronal CBCT slice

Image 1d: Sagittal CBCT slice

Case courtesy Prof. em. Dr. Daniel Buser, Clinic for oral surgery and somatology University Bern Prof. Dr. Michael Bornstein, Department of Oral Health &Medicine, University Center of Dental Medicine Basel UZB, University of Basel (Switzerland) and Honorary Professor in Oral and Maxillofacial Radiology, Facultyof Dentistry, The University of Hong Kong (Hong Kong SAR, China).
10Periodontics

Image 2a: Para-coronal view

Image 2b: Cross-sectional view

Image 2c: Axial view

Image 2d: Volumetric Rendered view

An asymptomatic 51-year-old male presents with numerous complaints associated with the mandibular anterior teeth includingunesthetic “black triangles” between the teeth, shrinking gums making the teeth look long, and “loose”.The patient reports anextensive dental history of periodontal bone loss and therapy. A 4 cm x 4 cm FOV at 0.08 nominal voxel size was acquired and paracoronal (a), cross-sectional (b), axial (c), and three dimensional volumetric rendered (d) images clearly show the presence of
calculus on the mandibular central incisors and generalized alveolar peri-circumferential radiographic bone loss extending to themiddle third of the root and beyond consistent with a diagnosis of Periodontitis Stage III, localized incisor pattern. The patient wasreferred to a periodontist for management.
Case courtesy, Drs William C. Scarfe and Gustavo Santaella, Louisville Kentucky

11Endodontics

Image 3a: Cross-sectional view

Image 3b: Para-sagittal view

An asymptomatic 27-year-old female presents immediately after fixed orthodontic appliance therapy with bitewing imaging
showing diffuse radiolucent increase in the middle third of the pulp chamber of the maxillary right canine compared to thecontralateral side. An 8 cm x 8 cm FOV at 0.25 nominal voxel size was acquired and a suspicious opacification of the pulp
canal of the maxillary right canine noted on axial (a) images. A 4 cm diameter “zoom reconstruction “was performed at anominal 0.08 mm voxel resolution centered on the maxillary canine. Cross-sectional (a), and para-sagittal (b) images clearly
identify dentin resorption on the mesio-palatal aspect of the root extending peri-circumferentially around the pulp canal andchamber and loss of the root continuity at the level of the cervical third of the root indicative of Class III invasive cervicalresorption. The patient was referred to an endodontist for management.

Image 4a: Axial view

Image 4b: Cross-sectional view

A 24-year-old female presents with vague tooth sensitivity in the maxillary right region. Her dental history reveals active
orthodontics more than 8 years previously involving extrusion of her right maxillary canine. Intraoral imaging is noncontributory. An 8 cm x 8 cm FOV at 0.25 nominal voxel size was acquired and a suspicious opacification of the pulp canal of themaxillary right canine noted. A 4 cm diameter “zoom reconstruction “was performed at a nominal 0.08 mm voxel resolution
centered on the maxillary canine and these indications are noted on axial (a) and magnified cross-sectional (b) images. This
image serves as a baseline for periodic, limited field, high resolution CBCT imaging to determine progression.
12

Case courtesy, Drs William C. Scarfe and Gustavo Santaella, Louisville KentuckyOral Surgery

Image 5a: MPR Panoramic view

Image 5b: Para-sagittal view

An asymptomatic 56 year-year-old male presents with a history of incidental discovery of possible mandibular pathology in theleft mandible on routine panoramic imaging taken 1 month previously. A 10 cm x 10 cm FOV at 0.25 nominal voxel size wasacquired, A reformatted MPR panoramic (a), and para-sagittal (b) images clearly show a single, well-defined, corticated, irregularly-shaped bilobular low density lesion anterior to the lingula and mandibular foramen within the left ascending ramus withextension inferiorly through the intramedullary bone to include the entire alveolus posterior to the distal root of the left mandibular
third molar. There is lingual cortical expansion but no perforation. The mandibular canal is intact throughout its course within theramus and the mandible. The patient was referred to an oral surgeon and excisional biopsy reveals an odontogenic keratocyst.
Case courtesy, Drs William C. Scarfe and Gustavo Santaella, Louisville Kentucky

Image 6a: Axial view

Image 6b: Volume Rendered view

Buccal bifurcation cyst. Buccal localization of the lesion with an extended resorption of the vestibular cortical plate
and a periosteal reaction.
Case courtesy, Prof. Dr. Reinhilde Jacobs, Center for Dentomaxillofacial Imaging, University Hospitals Leuven and Department of Imaging & Pathology, OMFS-IMPATH
Research Group, KU Leuven, Leuven, Belgium

13

14

Specifications

Trade Name:

3D Accuitomo

2000mm (6-9/16")

Type:EX1/2 F17Power Supply:

AC 100/ 110/ 120 V
AC 220/230/240 VAC

Power Consumption:

1800 mm (5-7/8")

MCT -11250 mm (49-1/4")

Model:

300 mm (11-3/4")

XYZ Slice View Tomographmax 2.0 kVA

Dimensions:
Main Unit:

W1,620 mm x D1,250 mm x H2,080 mm(63-3/4” x 49-1/4” x 82”)

Control Box:

1620 mm (63-3/4")

850 mm (33-1/2")

540 mm (21-1/4")

W100 mm x D40 mm x H115 mm(4” x 1-5/8” x 4-1/2”)

X-ray Head
Tube Voltage:

60-90 kV

Tube Current:

1-10 mA

2080 mm (82")

Approx. 400kg (Approx. 882lbs)
1500 mm (59")

Weight:(Max 8mA : Hi-Fi, Hi-Res Mode)
Focal Spot Size:

0.5

Exposure Time:

Std Mode: 17.5 / 9.0 sec(360º/180º)

Hi-Fi Mode : 30.8 / 15.8 sec
Hi-Res Mode : 30.8 / 15.8 sec
Hi-Speed Mode : 10.5 / 5.4 sec

Field of Viewø 40 x H40 mm
ø 60 x H60 mm
ø 80 x H80 mm
ø 100 x H50 mmø 100 x H100 mm
ø 140 x H50 mm
ø 140 x H100 mm
ø 170 x H120 mm

Voxel Size

80µm / 125µm / 160µm / 250µm* X-ray protection should be provided for the patient when X-rays are emitted.
* Design and specifications are subject to change without notification.

15

Diagnostic and Imaging Equipment
Treatment Units
Handpieces and Instruments
Endodontic System
Laser Equipment
Development and Manufacturing
J. MORITA MFG. CORP.
680 Higashihama Minami-cho, Fushimi-ku,
Kyoto 612-8533, Japan
T +81. (0)75. 611 2141, F +81. (0)75. 622 4595
Morita Global Website
www.morita.com
Distribution
J. MORITA CORP.
3-33-18 Tarumi-cho, Suita-shi, Osaka 564-8650, Japan
T +81. (0)6. 6380 1521, F +81. (0)6. 6380 0585
J. MORITA USA, INC.
9 Mason, lrvine CA 92618, USA
T +1. 949. 581 9600, F +1. 949. 581 8811
J. MORITA EUROPE GMBH
Justus-von-Liebig-Strasse 27b, 63128 Dietzenbach, Germany
T +49. (0)6074. 836 0, F +49. (0)6074. 836 299
MORITA DENTAL ASIA PTE. LTD.
150 Kampong Ampat#06-01A KA Centre, Singapore 368324
T +65. 6779. 4795, F +65. 6777. 2279
J. MORITA CORP. AUSTRALIA & NEW ZEALAND
Suite 2.05, 247 Coward Street, Mascot NSW 2020, Australia
T +61. (0)2. 9667 3555, F +61. (0)2. 9667 3577
J. MORITA CORP. MIDDLE EAST
4 Tag Al Roasaa, Apartment 902, Saba Pacha 21311 Alexandria, Egypt
T +20. (0)3. 58 222 94, F +20. (0)3. 58 222 96
J. MORITA CORP. INDIA
Filix Office No.908, L.B.S. Marg, Opp. Asian Paints, Bhandup (West), Mumbai 400078, India
T +91-22-2595-3482
J. MORITA MFG. CORP. INDONESIA
28F, DBS Bank Tower, Jl. Prof. Dr. Satrio Kav. 3-5, Jakarta 12940, Indonesia
T +62-21-2988-8332, F + 62-21-2988-8201
SIAMDENT CO., LTD.
71/10 Mu 5, Thakham, Bangpakong, Chachuengsao 24130, Thailand
T +66. 38. 573042, F +66. 38. 573043
www.siamdent.com

Subject to technical changes and errors.
Pub: No. DI129-B005-EN-4
JME EN 0221*0.5

Laboratory Devices
Educational and Training Systems
Auxiliaries