Diagnostic and Imaging Equipment

Specification

Treatment Units
Handpieces and Instruments
Endodontic System

Name:
Model:
Manufacturer:

Tri Auto ZX2TR-ZX2
J. MORITA MFG. CORP.

Functions:

Root Canal Measurement
OGP (Optimum Glide Path)OTR (Optimum Torque Reverse)
OAS (Optimum Apical Stop)
Auto Start / Stop
Auto Torque Reverse
Auto Apical Reverse / Stop
Auto Apical Slow Down
Auto Apical Torque Down
Auto Torque Slow Down

Accessories:

Tester, Spray Nozzle

Options:

LS Spray, Handpiece Holder,
External File Electrode (with cap)

Handpiece
Free Running Operation
Speed:Torque:Battery:
Weight:Battery Charger
Rated Input Voltage:
Rated Input Current:
Weight:AC Adapter
Rated Input Voltage:
Rated Input Frequency:
Rated Input Current:

Laser Equipment
100 ±20 – 1000 ±100 rpm
4 Ncm or more
Lithium-ion battery (DC 3.7 V)
Approx. 140 g (including batteryand contra angle)

DC 5 V
2.4 A
Approx. 280 g (AC adapter notincluded)

AC100 - 240 V
47 - 63 Hz
0.4 A

Development and Manufacturing
J. MORITA MFG. CORP.
680 Higashihama Minami-cho, Fushimi-ku,
Kyoto 612-8533, Japan
T +81. (0)75. 611 2141, F +81. (0)75. 622 4595
Morita Global Website
www.morita.com

Laboratory Devices
Educational and Training Systems
Auxiliaries

Distribution
J. MORITA CORP.
3-33-18 Tarumi-cho, Suita-shi, Osaka 564-8650, Japan
T +81. (0)6. 6380 1521, F +81. (0)6. 6380 0585
J. MORITA USA, INC.
9 Mason, lrvine CA 92618, USA
T +1. 949. 581 9600, F +1. 949. 581 8811
J. MORITA EUROPE GMBH
Justus-von-Liebig-Strasse 27a, 63128 Dietzenbach, Germany
T +49. (0)6074. 836 0, F +49. (0)6074. 836 299

Tri Auto ZX2MORITA DENTAL ASIA PTE. LTD.
3 Science Park Drive,#01-05 The Franklin Singapore Science Park1, Singapore 118223
T +65. 6779. 4795, F +65. 6777. 2279
Handpiece

Battery Charger

Disposable Cover(30 pcs.)

J. MORITA CORP. AUSTRALIA & NEW ZEALAND
Suite 2.05, 247 Coward Street, Mascot NSW 2020, Australia
T +61. (0)2. 9667 3555, F +61. (0)2. 9667 3577

Contrary ElectrodeJ. MORITA CORP. MIDDLE EAST
4 Tag Al Roasaa, Apartment 902, Saba Pacha 21311 Alexandria, Egypt
T +20. (0)3. 58 222 94, F +20. (0)3. 58 222 96

File Holder202

J. MORITA CORP. INDIA
Filix Office No.908, L.B.S. Marg, Opp. Asian Paints, Bhandup (West), Mumbai 400078, India
T +91-22-2595-3482
J. MORITA MFG. CORP. INDONESIA
28F, DBS Bank Tower, Jl. Prof. Dr. Satrio Kav. 3-5, Jakarta 12940, Indonesia
T +62-21-2988-8332, F + 62-21-2988-8201

Probe Cord72

SIAMDENT CO., LTD.
71/10 Mu 5, Thakham, Bangpakong, Chachuengsao 24130, Thailand
T +66. 38. 573042, F +66. 38. 573043
www.siamdent.com

3031

Industrial Design : f/p design gmbh

Ø 86(Unit: mm)

Catalog Design: f/p design, Germany
Subject to technical changes and errors.
17.03 TriAuto ZX2 Br En
JME EN BS 0218 *1.

Thinking ahead. Focused on life.

Apical Patency by Rotary Instrumentation

Apical Control

Now Possible with Rotary Files as Small as #10

Observe the File Tip Position with the Apex Locator Function

A Motor for Apical Patency,
Glide Path, and Shaping
Tri Auto ZX2 is a compact,cordless motor with abuilt-in apex locator. It canbe used for obtaining apicalpatency and creating a glide
path using rotary
instrumentation. Rotary
files even as Small as #10
may be used for theseprocedures. Responsive,the handpiece allows subtle
and delicate manipulation of
the file, corresponding tothe movements of anexperienced dentist's hand.
Tri Auto ZX2 also offers asafe and efficient method
for canal shaping.

W.L.

Apical Patency and Glide Path Using a Motor
OGP Function (Optimum Glide Path)
Using #20 or smaller files, the motor can be used to achieve
apical patency and create a glide path. It reproduces the subtle
and delicate finger movements of an experienced dentist.
Even curved or constricted canals can be treated faster andmore safely. Both NiTi files sized #20 or smaller, or stainlesssteel files sized #15 or smaller may be used.

Safer and Efficient Canal Shaping
OTR Function (Optimum Torque Reverse)
Depending on the load on the file, the motor alternates
between forward and reverse rotation with great sensitivity
to prevent jamming and file breakage. It is also superior in its
ability to follow the shape of even curved canals to reduce
ledges and over instrumentation. Tri Auto ZX2 is compatible
with standard NiTi files - no need to purchase specialty files.

Canal Shaping Procedure
Step 1

Step 2 / OGP

Canal shaping (Upper part) Patency

Clinical images are provided by Dr. Yoshi Terauchi / Dr. Toshihiro Ushikubo / Dr. Tai Gega

Step 3 / OGP

Step 4 / OTR

Glide path

Canal shaping

File tip position is monitored in real time.

File Tip Position Monitored
in Real Time
File location and canallength are monitored in realtime. Even if the working
length is altered by shaping,
the display accurately shows
the file position with respect
to the working length.

Automatic Stop Safety Function
The Tri Auto ZX2 motor is linked with the canal measurement function. Thisfeature makes endodontic treatment safer and more efficient. The position of thefile tip inside the canal is shown in the display, and the motor stops or reverses
rotation as soon as the file reaches the end of the working length to prevent
over instrumentation. Also the auto start and stop function automatically starts
the motor when the file is inserted into the canal and stops it when it isremoved.

OAS (Optimum Apical Stop) Function
When the file reaches the end of theworking length, it reverses slightly.
This reduces the possibility of filejamming and separation.

Precise Manual Canal Measurement
Canals can also be measured manually. A file holder is provided. The ratio of the impedances of two different
frequencies is calculated so that the location of the apex is determined with great accuracy regardless of whether thecanal is wet or dry. The file tip location is clearly shown in an easy to read meter display. The combination of tactile
feedback and the meter reading makes it easy to determine the location of the file tip. A beeping sound also acts as aguide to its location.