Veraviewepocs 2D

Thinking ahead. Focused on life.

Veraviewepocs 2D
The New Frontier of X-ray
MORITA‘s cutting edge technology produces 2D images of superior quality with highresolution and low X-ray exposure.
The Veraviewepocs 2D features a variety of specialized programs, such as theOrthoradial Panoramic projection, which reduces the overlapping of neighboring
teeth, and Shadow Reduction Panoramic projection, which reduces obstructing
shadows, as well as the AF (Auto-focus) function for accurate patient positioning.

23Panoramic

Conventional Image

Automatic Image Enhancer comparison
Automatic Image
Enhancement (AIE)Super High Quality Image
Veraviewepocs produces high resolution even in High Speed Mode. The resulting image offers superb density and contrast.
Digital Direct AE (Automatic Exposure) and Automatic Image Enhancement combine to give you the optimal image.

144 µmCPU96 µm

Fine High Speed Mode: pixel size 144 μm
Super Fine Mode: pixel size 96 μm

High ResolutionSensor

Automatic Image
Enhancement enhances thedetails that can be observedin areas which are either
extremely light or extremelydark. DDAE and AIE perform
a logarithmic conversion toproduce the highest quality
image possible.

X-ray Head

Digital Direct Automatic Exposure (DDAE)

Fine High Speed Mode:
DDAE adjusts the X-ray tube current (mA) dynamically by
At a pixel size of 144 μm, it produces superior high resolution images. detecting X-rays passing through the patient. This improves
the dynamic range, and, along with Automatic Exposure (AE),Super Fine Mode:
results in exceptionally clear images with the best possible
Produces an even better image with increased resolution.
contrast and density. The automatic exposure level can beadjusted to meet your individual requirements.

45PanoramicStandard Panoramic, Mag.: 1.3 x constant
The thick/specially-designed image layer accommodates all the possible variations of dentalarch shapes and sizes to produce extremely clear and sharp images.

TMJ 4 Views, Mag.: 1.3 x constant
Sharp, clear images of the TMJ are produced by aligning the angle of X-ray
penetration with the longitudinal axis of the mandibular condyle head.

Pedodontic Panoramic, Mag.: 1.3 x constant (Mag.: 1.6 x is also available)
For children or people with small jaws. The arm‘s rotation range isreduced, and thus reduces X-ray exposure.

Maxillary Sinus Panoramic, posterior Mag.: 1.5 x constant

67PanoramicStandard Panoramic,
Mag.: 1.3 x constant
The thick/specially-designed
image layer accommodates
all the possible variations ofdental arch shapes and sizes
to produce extremely clear
and sharp images.

Cephalometric

Shadow Reduction
Panoramic,
Mag.: 1.3 x constant(Mag.: 1.6 x is also available)
Produces images with lessmandibular ramus shadow.

Light emitter

Orthoradial Panoramic,
Mag.: 1.3 x constant(Mag.: 1.6 x is also available)
The perpendicular projectionof the X-ray reduces theamount of overlapping withemphasis on the maxillar
bicuspid region.

PSD sensor

AF Automatic Positioning

LAThe light beam sensor automatically positions the C-arm
without requiring the patient to move. It then measures thedistance to the patient's anterior teeth and AF automaticallymoves the C-arm into the optimal position. This creates
images with a high degree of reproducibility.

Patient health first: only 1/10* X-ray exposureWith only a tenth of the X-ray radiation, the radiation exposure is reduced significantly compared with conventional X-rays.

The semiconductor position detector (PSD sensor) measures distance withan extreme accuracy of 0.2 mm for high reproducibility. AF makes positioning
easy and precise.

High quality images with a wide dynamic range
Far more information about hard and soft tissue is received with just a single acquisition.
Fine High Speed CCD digital cephalometricHigh quality images in only 4.9 seconds.
Variable image processing techniques are used to generate an impressive grayscale range.

8* This comparison is made with the Veraviewepocs film-based system

9Specification
Veraviewepocs 2D
Panoramic

max. 1,020 mm (40-1/8")

2,355 mm (92-3/4")
min.1,055 mm (41-1/2")

EX-1: 96 mm (3-3/4")
EX-2: 70 mm (2-3/4")
115 mm (4-1/2")

40 mm (1-1/2")

max. 1,775 mm (69-7/8")

1,500 mm (59")

max. 1,380 mm (54-3/8")

max. 1,330 mm (52-3/8")

960 mm (38-3/4")

730 mm (28-3/4")

Panoramic Small Base

1,500 mm (59")

Panoramic

Panoramic/Cephalometric

Trade nameVeraviewepocs 2D

Model

Veraviewepocs 2D
Panoramic/Cephalometric

2DB

CassettePanPan/Ceph
EX-1: AC 120V 60 Hz, EX-2: 220/230/240 V 50/60 Hz

Power consumption

2,300 mm (90-1/2")

2,355 mm (92-3/4")

Tube voltage

60–80 kV

Tube current

1–10 mA

Effective focal spot

0.5 mmPanoramic
Exposure time

Fine high-speed mode approx. 7.4 seconds, Super fine mode approx. 15 seconds

Magnification ratio

1.3, 1.5, 1.6

Positioning
max. 1,775 mm (69-7/8")
min. 1,125 mm (44-1/4")

max. 1,380 mm (54-3/8")

max. 1,330 mm (52-3/8")
max. 1,020 mm (40-1/8")

2.0 kVA

X-ray generator

510 mm (20")

845 mm (33-1/4")

1,500 mm (59")

2DA

Input voltage

1,490 mm (58-5/8")

Electric motor and AF optical distance sensor

Cephalometric
Imaging area—

LA 225 x 254 mm, PA 225 x 203 mm

Magnification ratio—

1.1

W 1,020 x D 1,330 x H 2,355 mm(W 40-1/8“ x D 52-3/8“ x H 92-3/4“)W 2,000 x D 1,330 x H 2,355 mm(W 78-3/4“ x D 52-3/8“ x H 92-3/4“)Dimensions
Main unit
Control box
Installation area
Weight

10X550Type

W 70 x D 40 x H 115 mm (W 2-3/4“ x D 1-1/2“ x H 4-1/2“)
1.35 m2 (14.53 sf)

2.60 m2 (27.99 sf)

Approx. 190 kg (418 lb.)

Approx. 258 kg (568 lb.)

11

Diagnostic and Imaging Equipment
Treatment Units
Handpieces and Instruments
Endodontic System
Laser Equipment
Development and Manufacturing
J. MORITA MFG. CORP.
680 Higashihama Minami-cho, Fushimi-ku,
Kyoto 612-8533, Japan
T +81. (0)75. 611 2141, F +81. (0)75. 622 4595
Morita Global Website
www.morita.com
Distribution
J. MORITA CORP.
3-33-18 Tarumi-cho, Suita-shi, Osaka 564-8650, Japan
T +81. (0)6. 6380 1521, F +81. (0)6. 6380 0585
J. MORITA USA, INC.
9 Mason, lrvine CA 92618, USAT +1. 949. 581 9600, F +1. 949. 581 8811
J. MORITA EUROPE GMBH
Justus-von-Liebig-Strasse 27b, 63128 Dietzenbach, Germany
T +49. (0)6074. 836 0, F +49. (0)6074. 836 299
MORITA DENTAL ASIA PTE. LTD.
150 Kampong Ampat#06-01A KA Centre, Singapore 368324
T +65. 6779. 4795, F +65. 6777. 2279
J. MORITA CORP. AUSTRALIA & NEW ZEALAND
Suite 2.05, 247 Coward Street, Mascot NSW 2020, Australia
T +61. (0)2. 9667 3555, F +61. (0)2. 9667 3577
J. MORITA CORP. MIDDLE EAST
4 Tag Al Roasaa, Apartment 902, Saba Pacha 21311 Alexandria, Egypt
T +20. (0)3. 58 222 94, F +20. (0)3. 58 222 96
J. MORITA CORP. INDIA
Filix Office No.908, L.B.S. Marg, Opp. Asian Paints, Bhandup (West), Mumbai 400078, India
T +91-22-2595-3482
J. MORITA MFG. CORP. INDONESIA
28F, DBS Bank Tower, Jl. Prof. Dr. Satrio Kav. 3-5, Jakarta 12940, Indonesia
T +62-21-2988-8332, F + 62-21-2988-8201
SIAMDENT CO., LTD.
71/10 Mu 5, Thakham, Bangpakong, Chachuengsao 24130, Thailand
T +66. 38. 573042, F +66. 38. 573043
www.siamdent.com

Subject to technical changes and errors.
Pub: No. DI165-B00003-EN-3 JME EN0421

Laboratory Devices
Educational and Training Systems
Auxiliaries