AdvErL EVO
Gentle treatment for
wide-ranging disciplines

Thinking ahead. Focused on life.

AdvErL EVO
Versatility, efficiency and comfort in just one unit
With more than 20 years’ experience on the laser market, Morita is able to offer theAdvErL EVO Er:YAG laser, a sophisticated system which supports the economic
efficiency of dental practices in a wide variety of ways. AdvErL EVO operates on aminimally invasive basis and guarantees both safe and comfortable patient treatment.
In addition, the AdvErL EVO combines a large range of comfort features: Thescaler-formed handpiece ensures direct access to the preparation area and thenarrow application tips optimize the area with wide visibility.
Above all, the application tips are the unique feature of the AdvErL EVO andprovide water and air right to the tip and therefore precisely to aimed point ofemission. In this way, the laser develops its full effect precisely in the preparation
area and is suitable for a wide variety of indications. From peri-implantitis
treatment, periodontology, oral surgery, conservative therapy and endodontics –the Er:YAG laser has a very wide performance range.

2Er:YAG laser technology
for a wide variety of treatments

Less painful
The wavelength of the Er:YAG laser
is the most suitable for dental treat­ment because it is readily absorbed bywater. Therefore it efficiently vaporizes
hu­man tissue that has a high watercontent and effects only the surface ofthe tissues. It also produces very little
heat which reduces pain to a minimum.

Less trauma for biological tissues
Unlike CO 2 and Nd:YAG lasers, vapo­
rization is concentrated at the surfaceof the tissue so the energy does not
penetrate and damage deeper layersof tissue. It is also far less likely topro­duce cracks in the enamel, and it
does not disperse energy to the areasurrounding the irradiation target.

Er : YAG laser

Wide variety of applications
The wide array of tip options enables
this laser to perform both hard and soft
tissue procedures.
CO 2 laser

Periodontal treatment

Soft tissue treatment

Hard tissue treatment(incision, excision, vaporization, ablation and coagulation)(incision, excision, vaporization, ablation and coagulation)(ablation, vaporization)

Removal of subgingival calculi
Laser soft tissue curettage

Gingival incision and excision
Hemostasis and coagulation
Frenectomy and frenotomy

Class I, II, III, IV and V cavity preparation
Caries removal

Scaling4

Nd : YAG laser

Gingival incision and excision

Caries removal

Clinical indications

Hard tissue indications
Caries removal
Surface ablation of wedge
shaped defect
Periodontal indications
Irradiation to the periodontal pocket
Scaling
Periodontal curettage
Gingivoplasty
Gingival flap operation
Soft tissue indications
Frenectomy and frenotomy
Gingival incision and excisionCoagulation of stomatitis
Removal of pigmentation

Endodontic indications
L aser root canal disinfection after
endodontic treatment
Flap preparation – incision of soft
tissue to prepare a flap and expose
the bone
Cutting bone to prepare a window
access to the apex (apices) of theroot(s)Apicoectomy
Root end preparation for re-fill with
bio-ceramic materials
Removal of pathological tissues(e.g. cysts, neoplasm or abscess)and hyperplastic tissues (e.g. granulation tissue) from around the apex

5

Smooth handling,easy navigation

Intuitive interfaceThe display is large, easy to read, andhas an intuitive design. All settings canbe easily confirmed at a glance. Thereare 20 pre-programmed settings thatcan be easily retrieved.

6Usage log saved on flash memory
A usage log is saved on a flash memory
stick that can easily be connected to acomputer for review.

approx. 600 mm
handpiece movement range

Elegant and compact
The unit is lightweight and can easily be
moved with large castors and handles onboth the front and back. These convenientfeatures also allow the unit to be slightlyrepositioned during treatment if needed.
To install the laser, simply plug it in.
The water and air systems are all built
in eliminating the need for complicatedconnections.

Scaler-formed handpieceThis special formed handpiece offers abetter access and a direct view to the
preparation area.
Easy handling and operation
The flexible optical transmission cableand swing arm make the handpieceseem nearly weightless and allows thedentist to concentrate on the treatment
procedure.

Adjustable handpiece hanger
The hanger is attached to the unit witha magnet so that it remains secure. The
position of the handpiece hanger can beadjusted to suit the individual dentist.
Infection control
The handpiece hanger can be autoclavedfor enhanced infection control.

7

Revolutionary new treatment
for peri-implantitis

Peri-implantitis is a disease that currently poses a very challenging clinical problem. Research has shownno definitive, evidence based treatments that lead to a permanentand predictable outcome.
Now, under the leadership of Dr. Atsuhiko
Yamamoto of the Japan Institute forAdvanced Dental Studies, Perio-Implant
Hospital AUTIS, we have developed
an effective therapy that has alreadyachieved great success in Japan during
the last 8 years. Using our AdvErL Evo
Er:YAG laser, the tissue will regenerate for long term and the bacteria isremoved permanently by the utilizationof micro-explosions produced by thelaser's energy. It's an extremely low
heat treatment which doesn't interferewith osseointegration and removes thecontaminated titanium oxidized layer.
The micro-explosions are the key toour epoch-making treatment method.
Micro-explosions exist when the laser
energy is absorbed by water, the vol-8

ume suddenly expands from 800 to1,000 times. We have found a way toexploit these micro-explosions so thatboth the contaminated matter on theimplant surface and the layer of the contaminated oxidized titanium is removed.
With this method, the surface is not
scraped off; the titanium is peeled off.
So the unique property of using ourAdvErL Evo Er:YAG laser is its wave­
length; compared to the wavelengthof other lasers, the absorbtion of waterby our system is much greater.

Clinical case study

A female patient complained of swellingaround an implant. An X-ray examination
showed consi­­derable transparency aroundthe implant in the region of tooth 19. The
superstructure was re­moved and it wasobserved that the implant was not loose.
An examination deter­mined that this wasa CIST class D case.

After opening a flap, bone loss andalso gra­­nu­lation tissue were observed.
The granu­lation tissue was removed
with Er:YAG laser irradiation. The key toremoving this tissue is not to irradiate
the tissue itself, but to irradiate the layer
where the contaminated and healthy
tissue meet.

X-rays taken before treatment, 3 months
after treatment and 2 years later. The im­plant has stabilized.

The CT and volume rendered images
show that bone growth has been satis­
factory and the implant has stabilized.

After removing the contaminated tissue,
the oxidized layer on the implant was ster­i­lized and peeled off.

9

Tip options

A total of 21 tips are available
for a wide range of treatments.

C 40 0F

C60 0F

C80 0F

CS600FP40 0TP60 0TP40 0FL

PS40 0T

PS60 0T

PS80 0T

PS40 0TSPS60 0TS

PS80 0TS

PSM60 0T

CF60 0

S60 0T

Brush

R135T

R20 0T

R30 0T

R60 0T

Standard tips are indicated by agreen square. The R series tips
require an additional R-Attachment.

Marking Tool:

CCFPPS
R
Brush
S

10Standard Tip
Handpiece Type R is necessary
Caries Tip
Corner/Curve flat tip
Perio tip
Pocket surgery tip
Root tip
Hyper sensitivy
Surgery tip

Specifications

AdvErL EVO

Model

MEY-1-A

Power rating

AC 100 V – 240 V ±10% 50/60 Hz

Power consumption

1.5 kVA (at maximum laser output)

Laser classification

Class 4 (Er:YAG), Class 2 (aiming beam)Type of laser

Er:YAGWavelength

2,940 nmType of operation

Pulse

Laser energy

30 ~ 400 mJ at 1 ~ 10 PPS
30 ~ 170 mJ at 20 PPS
30 ~ 80 mJ at 25 PPS

Pulse repetition rate

1, 3.3, 5, 10, 20, 25 PPS

Aiming beam

LD (red)Aiming beam wavelength

650 ±15 nm

Type of protection against electric shock

Class I

Degree of protection against electric shock

Type B

Cooling method

Water cooling (built-in reserve)

Outer dimensions(W)246 × (D)585 × (H)732 mm

WeightApprox. 49 kg

732

1715

Name246585Dimensions（mm）

11

Diagnostic and Imaging Equipment
Treatment Units
Handpieces and Instruments
Endodontic Systems
Laser Equipment
Laboratory Devices

Distribution
J. Morita Europe GmbH
Justus-von-Liebig-Str. 27a63128 Dietzenbach
Germany
T +49. 6074. 836 0, F +49. 6074. 836 299
www.morita.com/europe
Development and Manufacturing
J. MORITA MFG. CORP.
680 Higashihama Minami-cho, Fushimi-ku
Kyoto 612-8533
JapanT +81. 75. 611 2141, F +81. 75. 622 4595
www.morita.com

Subject to technical changes and errors.
JME EN PUB 2029 1216 *3.