VeraviewIC5 HD

Thinking ahead. Focused on life.

Veraview IC5 HD
High Definition Panoramic Imaging
Industry Leading Exposure Time
Extremely fast 5.5 second panoramic with significantly lower dosage

High Definition
Improved definition of 96µm (10 second scan) for the ultimate high definition image

Class-leading clarity
DDAE (Digital Direct Auto Exposure) and AIE (Auto Image Enhancement) functions provide
sharp and clear images

Quick and Easy Patient Positioning
With integrated triple laser beam alignment

Fully Automatic Simplified Operation
Simply select the desired program: Panoramic, TMJ or PEDO. In addition, no parameter setting
is required

Digital Technology
No film or film development are required

Low Power Consumption
Lightweight and Compact Design23

Select from two levels of speed and resolution

The combination of DDAE (Digital Direct Auto Exposure) and AIE (auto imageenhancement) produces high quality images with optimum contrast and uniform density.

Depending on the diagnostic need, you can take a quick, 5.5 second panoramic
for the lowest dose, or a 10 second scan for the ultimate, high resolution image.

Digital Direct Automatic Exposure (DDAE)
CPUDDAE controls the X-ray tube current (mA) simultaneously bydetecting X-rays passing through the patient. This improves
the dynamic range, and, along with Automatic Exposure (AE),results in exceptionally clear images with the best possible
contrast and even density. The automatic exposure level canbe adjusted to meet your individual requirements.

High Speed
With an industry leading
exposure time of 5.5
seconds, the Veraview IC5
is not only significantly
faster than conventionalfilm-based panoramicsystems, but also requires
much less radiation.

Sensor

X-ray HeadHigh Speed: only 5.5 seconds, 192µm, 1/6** X-Ray exposure
Before auto image enhancement (AIE)High Definition
Pixel size is reduced 25%
compared with the former
model, so it produces
superior images of a higherresolution.

Automatic Image Enhancer comparison
Automatic Image Enhancer(AIE)
AIE is a special software
function that optimizes
density throughout an entire
image, and highlights thebrightness level of detailed
areas, to create images withan extremely high degree ofclarity. DDAE and AIE perform
a logarithmic conversion toproduce the highest quality
image possible.

192µｍ
144µｍ
96µｍ

High Definition: 10 second exposure, 96 µm

High Speed: 5.5 seconds (192µm)
High Definition: 10 seconds (96µm)

After auto image enhancement (AIE)* as of January 2010**Compared to a standard panorama image using
film and made by Morita’s Veraviewepocs (16seconds, 75kV, & 9mA)

45

Specifications

Power consumption:
Dimensions
Main unit:
Weight:X-ray generator
Tube voltage:
Tube current:
Effective focal spot:

Pedodontic Panoramic at
Reduced Radiation(Unit: mm)

970
680

W 890 x D 970 x H 2350
Approx. 110 kg

60-70kV (depending on exposure mode)
1-7.5mA (depending on exposure mode)
0.5 mm4055

Panoramic image
Exposure time: 	High speed mode: Approx. 5.5 sec.
High definition mode: Approx. 10 sec.

120

Imaging programs:	Standard Panoramic
Pedodontic Panoramic
TMJ Quadruple Image- Always have patients wear X-ray protective gear.

770660Max. 890

1500

Max. 970

For examinations of children
or people with small jaws,
the rotation range of the arm
is diminished, thereby further
reducing radiation exposure.

Veraview IC5 HD
XDP1
EX-1: AC 100V/115V/120V 50/60 Hz
EX-2: AC 220/230/240V 50/60 Hz
0.85 kVA

2350 (Option: 2180)

Clear, sharp images with awide image layer. The thick,specifically designed image
layer accommodates all possible
variations of dental arch shapes
and sizes to produce extremely
clear and sharp images.

Trade name:
Model:
Input voltage:

Max. 1830 (Option: max 1660)
Min. 1045 (Option: min. 960)

Standard Panoramic

115

Panoramic

TMJ Quadruple Exposures:for Open and Closed Condyle
Views
Four separate shots of condyle
fit onto one image. In each case,two sets of rotation for left andright condyle head for open andclosed condition.

Clinical images provided by: Dr. Bruno
A zevedo, Director of Radiology andImaging Sciences, University of Louisville
School of Dentistr y and Kitasenju Radist
Dental Clinic, i -View Imaging Center,Japan.

6

1500

Small BaseStandard Base
7

Diagnostic and Imaging Equipment
Treatment Units
Handpieces and Instruments
Endodontic System
Laser Equipment
Development and Manufacturing
J. MORITA MFG. CORP.
680 Higashihama Minami-cho, Fushimi-ku,
Kyoto 612-8533, Japan
T +81. (0)75. 611 2141, F +81. (0)75. 622 4595
Morita Global Website
www.morita.com
Distribution
J. MORITA CORP.
3-33-18 Tarumi-cho, Suita-shi, Osaka 564-8650, Japan
T +81. (0)6. 6380 1521, F +81. (0)6. 6380 0585
J. MORITA USA, INC.
9 Mason, lrvine CA 92618, USA
T +1. 949. 581 9600, F +1. 949. 581 8811
J. MORITA EUROPE GMBH
Justus-von-Liebig-Strasse 27a, 63128 Dietzenbach, Germany
T +49. (0)6074. 836 0, F +49. (0)6074. 836 299
MORITA DENTAL ASIA PTE. LTD.
150 Kampong Ampat#06-01A KA Centre, Singapore 368324T +65. 6779. 4795, F +65. 6777. 2279
J. MORITA CORP. AUSTRALIA & NEW ZEALAND
Suite 2.05, 247 Coward Street, Mascot NSW 2020, Australia
T +61. (0)2. 9667 3555, F +61. (0)2. 9667 3577
J. MORITA CORP. MIDDLE EAST
4 Tag Al Roasaa, Apartment 902, Saba Pacha 21311 Alexandria, Egypt
T +20. (0)3. 58 222 94, F +20. (0)3. 58 222 96
J. MORITA CORP. INDIA
Filix Office No.908, L.B.S. Marg, Opp. Asian Paints, Bhandup (West), Mumbai 400078, India
T +91-22-2595-3482
J. MORITA MFG. CORP. INDONESIA
28F, DBS Bank Tower, Jl. Prof. Dr. Satrio Kav. 3-5, Jakarta 12940, Indonesia
T +62-21-2988-8332, F + 62-21-2988-8201
SIAMDENT CO., LTD.
71/10 Mu 5, Thakham, Bangpakong, Chachuengsao 24130, Thailand
T +66. 38. 573042, F +66. 38. 573043
www.siamdent.com

Subject to technical changes and errors.
Pub: No. DI165-B00001-EN-1
JME EN 0919 *1.

Laboratory Devices
Educational and Training Systems
Auxiliaries