DBK213

Screw-Terminal & Expansion Card ModuleFeatures• Provides screw-terminal connections
for analog, digital, and counter signals• Includes P1, P2, P3, and P4 connectors• Accommodates any three analog ordigital DBK cards
The DBK213 screw-terminal and expansion module provides screw-­
terminal access plus expansion slots for­DaqBook/2000 Series, DaqBoard/2000­Series, DaqLab/2000 Series, ­DaqScan/2000­Series, and LogBook/300 Series products.
The top of the DBK213 is easily removed
to reveal screw-terminal connections thatprovide access to all analog and digitalI/O from the host data acquisition ­module.
Also included with the DBK213 are three
expansion slots that can accommodate any
DBK analog or digital expansion and signal
conditioning cards.

The DBK213 provides screw-terminal signal connections and three DBK expansion and signal
conditioning card slots (cards shown are not included)

The front of the DBK213 contains slots
through which all of the signal I/O wires
from the screw terminals can be routed.
The rear of the DBK213 contains P1,P2, P3, and P4 connectors to connect
the screw-­terminal board to the dataacquisition host.

DBK213 System Examples
FrontBack

Notebook or Desktop PC

Ethernet

When used with DaqBoard/2000 Seriesboards, only a single CA-195 cable is required from the DaqBoard to the DBK213
to provide access to all 100 signals of theDaqBoard via the DBK213’s 100 screwterminal connections. In ­addition, P1 andP2 connectors on the DBK213 provide
easy connection to DBK expansion cards
installed into any of the 3 expansion slots,or to an external DBK expansion module(see diagram).
When used with other Daq or LogBookproducts, the DBK213 attaches to the P1,
P2, and P3 connectors on the data acquisition host using a separate CA-255 cablefor each port.

DaqBook/2001
Ethernet 16-bit, 200-kHz
data acquisition system

P1

P2P3

CA-255-2T (3)

DBK213
CA-37-3
CA-255-4T

DBK85
16-channel differential
voltage input module
DBK213
Back View

CA-195
P4P4

DaqBoard/2000
16-bit, 200-kHz PCI data acquisitionboard with 16 analog inputs,
40 digital I/O, and 2 analog outputs

CA-37-3

Measurement Computing(508) 946-5100

1

info@mccdaq.com

mccdaq.com

DBK213

Specifications & Ordering Information
SpecificationsCable (optional): CA-195, CA-37-X, or CA-255-X
Operating Environment
Temperature: -30 to 70 ˚C
Relative Humidity: 95% RH, non-condensingConnectors
P4: 100-pin connector provides for connection to DaqBoard/2000 Series P4
connector via a CA-195 cable
P1: One P1 (DB37) connector provides for analog expansion or DaqBook/2000
Series, DaqLab/2000 Series, DaqScan/2000 Series, and LogBook/300 Seriesconnection
P2: One P2 (DB37) connector provides for digital expansion or DaqBook/2000
Series, DaqLab/2000 Series, DaqScan/2000 Series, and LogBook/300 Seriesconnection
P3: One P3 (DB37) connector provides for counter and analog output connection or DaqBook/2000 Series, DaqLab/2000 Series, DaqScan/2000 Series,
and LogBook/300 ­Series connection
Screw Terminals: 12 banks of 10 connector blocks
Dimensions: 285 mm W x 220 mm D x 45 mm H; (11” x 8.5” x 2.7”)
Weight: 1.45 kg (3.2 lbs)

Ordering Information

Description
Screw-terminal and expansion card module for DaqBook/2000 Series,
DaqBoard/2000 Series, DaqLab/2000 Series, DaqScan/2000 Series,
and LogBook/300 Series; accommodates up to three DBK
signal conditioning and expansion cards

DBK213 shown with cover removed, providing access to all screw
terminals

Part No.

DBK213

Accessories & Cables
Rack-mount kit
100-conductor expansion cable, mates with P4
on the DaqBoard/2000 Series		 3 ft. expansion cable		 6 ft. expansion cable
Same as CA-195 with CE compliance; 3 ft.
Molded T expansion cable; 2 in.
Molded T expansion cable; 4 in.
Ribbon cable, where x is the number of DBK devices attached

RackDBK4
CA-195
CA-195-6
CA-209CA-255-2T
CA-255-4T
CA-37-x

Product Compatibility

Rear view of DBK213 showing all connections to the A/D host via
P1, P2, P3, and P4✔ LogBook✔ DaqBook✔ DaqLab✔ DaqScan✔ DaqBoard/2000 Series

Measurement Computing(508) 946-5100

2

info@mccdaq.com

mccdaq.com