Pressure Comparison Pump LPP 60-T
Generang pressure -0,95...+60 bar (870 psi)

LPP 60-T
Rel. 20191007

Pressure Comparison Test Pump LR-Cal LPP 60-T, pneumacGeneraon of test pressure up to +60 bar (870 psi), switchable toGeneraon of vacuum to -0.95 bar (-28 inHg)
The Calibration Test Pump LR-Cal LPP 60-T is used to generatepressures and vacuum for checking, adjusting and calibrating mechanicaland electronic pressure measuring instruments by comparative measurements.
These pressure tests may be carried out in laboratories, workshop or on site at themeasuring point. If the instrument under test and a sufficiently accurate reference
measuring instrument (e.g. Electronic Pressure Calibrator LR-Cal LPC 300 or
LR-Cal LPC 200, ±0.025% FS) are connected to the LR-Cal LPP 60-T, thesame pressure / vacuum is applied to the two instruments when
the pump is operated.

Despite its compact dimensions, the calibration test pump
LR-Cal LPP 60-T is easy to operate and allows for exact generationof the required test pressures. A change-over switch enables thegeneration of vacuum as well. The LR-Cal LPP 60-T is fitted with afine adjustment valve for the precise adjustment of pressures. The reference instrument is screwed directly
on to the top of the pump and the unit under test is connected by means of a connection tube incorporating
an adapter 1/4" BSP rotating female (optional 1/4" NPT female) port.

Specification:
Pressure generation:
Vacuum generation:
Pressure ports:

Volume per stroke:
Materials:
Fine adjustment:
Dimension:
Weight:0...60 bar = 0...870 psi (or equivalent), switchable to0...- 0,95 bar = 0...-28 inHG (or equivalent)
1/2" BSP female rotating (with seal) for reference instrument on the top of the pump(Type LPP-60-T-N14: 1/4" NPT female, type LPP-60-T-G14: 1/4" BSP female rotating)
1/4" BSP female for unit under test (at connection tube)(Type LPP-60-T-N14: 1/4" NPT female)appr. 11 cm³
anodized aluminum, brass (chromium plated)
Fine adjustment valve
appr. 150 x 300 x 215 mm
appr. 2.55 kg

Included in standard supply:• Calibration test pump LR-Cal LPP 60-T
• Connection tube with female rotating pressure port
• Operating manualOptional accessories:• Set of adapters and seals for unit under test
• Carrying case with custom foams• Tool for a simple operation of the adjustment valve(LPP-VOLUMEN-TOOL)

Recommended Reference Instruments:
••
••••
••
•

Accuracy ±0.025% FS:
Accuracy ±0.025% FS:
Accuracy ±0.050% FS:
Accuracy ±0.1% FS:
Accuracy ±0.2% FS:
Accuracy ±0.125% F
Accuracy ±0.25% F
Accuracy ±0.6& FS:
Accuracy ±1.0% FS:

Electronic Process Calibrator LPC 300
Electronic Pressure Calibrator LPC 200
Electronic Pressure Calibrator TLDMM
Digital Test Pressure Gauge TLDMM-A01
Digital Test Pressure Gauge TLDMM-A02
Digital Test Pressure Gauge LDM70-E25
Digital Test Pressure Gauge LDM70-K50
Analogue Pressure Gauge
LPP-MANO-K06
Analogue Pressure Gauge
LPP-MANO

datasheet “LPC 300"
datasheet “LPC 200"
datasheet “TLDMM 2.0”
datasheet “LDM 80"
datasheet “LDM 80"
datasheet “LMD 70"
datasheet “LDM 70"
see reverse page
see reverse page

DRUCK & TEMPERATUR Leitenberger GmbH • Bahnhofstr. 33 • D-72138 Kirchentellinsfurt • GERMANY
Tel. +49 (0) 7121-90920-0 • Fax +49 (0) 7121-90920-99 • DT-Info@Leitenberger.de • www.LR-Cal.net • www.druck-temperatur.de1

Pressure Comparison Pump LPP 60-T

LPP 60-T Generang pressure -0,95...+60 bar (870 psi)
Order-Codes:
Description

Order CodeLR-Cal LPP 60-T Calibration Test Pump -0,95...+60 bar (-28 in Hg...+870 psi)

LPP-60-T

NOTE: if you dismount the reference connector, you get 3/8" BSP female portditto, but reference port 1/4" BSP female rotating

LPP-60-T-G14

ditto, but reference port and test port 1/4" NPT female

LPP-60-T-N14

Set of BSP adapters and set of seals 1/8" BSP, 3/8" BSP and 1/2" BSP

LPP-ADAPTER-BSP

Set of metric adapters and set of seals M12x1,5M20x1,5 and Minimess®

LPP-ADAPTER-M

Set of NPT adapters 1/8" NPT , 1/4" NPT, 3/8" NPT, 1/2" NPT

LPP-ADAPTER-NPT

Tool to simplify operationof the fine adjustment valve

LPP-VOLUMEN-TOOL

Carrying case with foams

LPP-60-T-KOFFER

LPP-MANO-K06 analogue reference gauge DS 100±0.6% FS, all stainless steel, fine graduation dial. Incl. certificate
Available ranges: 0...4 bar/PSI; 0...25 bar/PSI; 0...40 bar/PSI;-1...0 bar/inHg; -1...+39 bar/PSI

LPP-MANO-....-K06

LPP-MANO analogue reference gauges DS 63±1% FS, fine graduation dial.
Available ranges: 0...2 bar/PSI; 0...11 bar/PSI; 0...25 bar/PSI;
0...40 bar/PSI; -1...0 bar/inHg; -1...+39 bar/PSI

LPP-MANO-....

Spare part: T est tube with 1/4" BSP female port

LPP-SCHLAUCH-S-0050

Spare part: Fine adjustment valve with relief valve

LPP-VENTIL

Service Kit for LR-Cal LPP 60-T (O-rings, seals, etc.)

LPP-WARTUNG

NOTE:
To mount the reference instruments LR-Cal LPC 300,
LPC 200, TLDMM 2.0, LDM 80, LDM 70
or LPP-MANO you do not need any
adapter if you use the calibration handpump
type LR-Cal LPP 60-T, order-code LPP-60-T .

(1) pressure port for mounting reference pressure instrument(2) fine volume adjustment, for generating high pressure appr.
40...60 bar and fine pressure adjustment(3) pressure relief valve(4) Switch pressure generation / vacuum generation(5) pressure port for test item(6) pump to generate pressure up to appr. 40 bar

2