MODEL 3058A ACCELEROMETERSPECIFICATIONS


Piezoresistive MEMS
DC Response, Gas Damped
Screw Mounted
Integral Temp Compensation

The Model 3058A is a silicon MEMS accelerometer withintegral temperature compensation. The accelerometer ispackaged on a ceramic substrate with a metal bracket which
can be used to bolt the sensor to the mounting location. Theaccelerometer is offered in ranges from ±2g to ±100g range
and provides a flat frequency response to minimum 1500Hz.
The silicon MEMS sensor is gas damped and incorporates
over-range stops for high-g shock protection.

dimensions

For a similar accelerometer designed for adhesivemounting, see the model 3052A.

FEATURESBolt Mounted±1.0% Non-Linearity
0 to +50°C Temp Compensation
Built-in Over-range Stops
Low Power Consumption

APPLICATIONSSENSOR SOLUTIONS /// Model 3058A Rev A

Vibration & Shock Monitoring
Motion Control
Impact & Shock Testing
Transportation Measurements
Embedded Applications
Machinery

07/2017

Page 1

MODEL 3058A ACCELEROMETERPERFORMANCE SPECIFICATIONS
All values are typical at +24°C, 80Hz and 5Vdc excitation unless otherwise stated. Measurement Specialties reserves the right to update
and change these specifications without notice.

1Parameters
DYNAMIC
Range (g)Sensitivity (mV/g)1
Frequency Response (Hz)Natural Frequency (Hz)Non-Linearity (%FSO)
Transverse Sensitivity (%)
Damping Ratio
Shock Limit (g)±28.0-16.0
0-150
700
±1.0
<30.7
3000

ELECTRICAL
Zero Acceleration Output (mV)
Excitation Voltage (Vdc)
Input Impedance (Ω)
Output Impedance (Ω)
Insulation Resistance (MΩ)
Residual Noise (µV RMS)
Ground Isolation±22.7 to 12
1200-6500
1200-6500>10010Isolated from Mounting Surface

Differential

ENVIRONMENTAL
Thermal Zero Shift (%FSO/°C)
Thermal Sensitivity Shift (%/°C)
Operating Temperature (°C)
Compensated Temperature (°C)
Storage Temperature (°C)
Humidity±0.060±0.060-40 to +125
0 to +50-40 to +125
Epoxy Sealed, IP61

0 to +50°C
0 to +50°C

PHYSICAL
Case Material
Weight (grams)
Mounting
Mounting Torque

Aluminum Flange, Ceramic Cover
4.5
2x #4-40 Mounting Screws
6 lb-in (0.7 N-m)

Notes
±54.8-7.20-250
800±1.0<30.7
3000±102.4-3.60-350
1000±1.0<30.7
3000±201.2-1.80-550
1500±1.0
<30.7
3000

±500.48-0.72
0-10004000±1.0
<30.6
5000

±1000.24-0.360-1300
6000
±1.0
<30.55000@5Vdc Excitation±5%<1 Typical@50VdcMaximum

Output is ratiometric to excitation voltage

Calibration supplied:

CS-SENS-0100

NIST Traceable Amplitude Calibration at 100Hz

Optional accessories:

121
140A

Three Channel DC Signal Conditioner Amplifier
Auto-Zero Inline Amplifier

The information in this sheet has been carefully reviewed and is believed to be accurate; however, no responsibility is assumed for inaccuracies. Furthermore, thisinformation does not convey to the purchaser of such devices any license under the patent rights to the manufacturer. Measurement Specialties, Inc. reserves the rightto make changes without further notice to any product herein. Measurement Specialties, Inc. makes no warranty, representation or guarantee regarding the suitabilityof its product for any particular purpose, nor does Measurement Specialties, Inc. assume any liability arising out of the application or use of any product or circuit andspecifically disclaims any and all liability, including without limitation consequential or incidental damages. Typical parameters can and do vary in different applications.
All operating parameters must be validated for each customer application by customer’s technical experts. Measurement Specialties, Inc. does not convey any licenseunder its patent rights nor the rights of others.

SENSOR SOLUTIONS /// Model 3058A Rev A

07/2017

Page 2

MODEL 3058A ACCELEROMETER

ORDERING INFOPART NUMBERING

Model Number+Range+Electrical Connection

3058A-GGG-P
l l_____Electrical Connection (P=pins)l___________Range (010 is 10g)

Example: 3058A-010-P
Model 3058A, 10g, Pins

TE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred to herein
might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard
Terms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuseof the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies

All Rights Reserved.

SENSOR SOLUTIONS /// Model 3058A Rev A

07/2017

Page 3