Pressure transmitter

OEM Relative and absolutepressure transmitter
Type 515
Type 515 pressure transmitter with cable connection meetsthe highest demands for industry and vehicle manufacturing
applications.
The compact and robust mechanical design incorporating
protection standard IP69K allows the use in stringent
conditions.

Pressure range-1 ... 0 – 600 bar+ Compact, rugged construction for highest operationalreliability+ Negligible temperature influence on accuracy+ Excellent EMC capacity

Pressure and flow

Huba Control Type 515 | Technical data subject to change | Edition 09/2020

1/6

Technical overview
Pressure range 1)
Relative
Absolute-1 ... 0 – 600 bar
0 ... 25 bar

Operating conditions
MediumTemperature

Tolerable overload / Rupture pressure 2)
Materials
Case
Materials in contact with the medium

FPM
EPDM
NBR
FPM spec.
Ambient<6>6Liquids-15 ... +125 ºC-25 ... +125 ºC-25 ... +85 ºC-40 ... +150 ºC
max. +85 ºC
3.0 x fs
2.5 x fs (max. 900 bar)Pressure connection
SensorMedia stop sytsem
Sealing material

Stainless steel 1.4305 / AISI 303
Stainless steel 1.4305 / AISI 303
Ceramic Al2O3 (96%)
PPS
FPM, EPDM, NBR, FPM spec.

Media stop system
Patented media stop system to prevent media egress when exeeding rupture pressure range (> 40 bar nominal value).
Electrical overview
2 wire3 wirePolarity reversal protection
Insulation voltage

OutputPower supply
Load
Current consumption 3)-8V
4 ... 20 mA
8.0 ... 33 VDC< supply voltage[Ohm]< 20 mA
0.02 A
0 ... 5 V
8.0 ... 33 VDC>10 kOhm / < 100 nF< 4 mA
1 ... 6 V
8.0 ... 33 VDC>10 kOhm / < 100 nF< 4 mA
0 ... 10 V
11.4 ... 33 VDC>10 kOhm / < 100 nF< 4 mA
0 ... 10 V
24 VAC ±15%
>10 kOhm / < 100 nF< 4 mAratiom. 10 ... 90%
5 VDC ±5%
>10 kOhm / < 100 nF< 4 mA
Short circuit proof and protected against polarity reversal. Each connection is protected against crossover up to max. supply voltage.
standard
500 VDC
optional
1000 VDC

Dynamic response
Response time
Load cycle< 2 ms, typ. 1 ms
< 100 Hz

Protection standard
IP 69K
Electrical connection
Cable PUR 1.5 m
Pressure connection
Inside thread

G ¼, with O-Ring sealG ¼, sealed at back, DIN 3852-E
G ½, sealed at front
G ½, sealed at back and manometer (combi)¼, -18 NPT
R ¼, EN 10226
M12x1.5 sealed at back, DIN 3852-E
M14x1.5 sealed at back, DIN 3852-E

Outside thread

Installation arrangement
Unrestricted
Tests / Admissions
Electromagnetic compatibility
Shock acc. IEC 60068-2-27
Constant shock acc. IEC 60068-2-29
Vibration acc. IEC 60068-2-6

CE conformity acc. EN 61326-2-3
40 g, 6 ms half sine wave, all 3 directions. Free fall from 2 m on concrete (6x)40 g for 6 ms, 1000 x all 3 directions
25 g, 2 ... 2000 Hz with amplitude ± 15 mm, 1 Octave/min. all 3 directions, 50 constant load

WeightWith thread G ½
All other versions~ 225 g~ 180 g

Packaging (Please state on order)
Single packaging in cardboard
Multiple packaging in cardboard per 25 pcs.

Accuracy
Parameter

Tolerance zero point
Tolerance full scale
Resolution
Total of linearity,hysteresis and repeatability
Long term stability acc. DIN EN 60770
TC zero point 4)
TC sensitivity 4)

1)

2/6

Other pressure range on request

Pressure and flowmax.
max.
max.
max.
max.

2)

Unit

Test conditions:% fs
% fs
% fs

± 0.3
± 0.3
0.1

% fs% fs
% fs/10K% fs/10K

± 0.3
± 0.5
± 0.150/-0.3

Other overload and Rupture pressure on request3)

25 ºC, 45% RH, power supply 24 VDC
TC z.p. / TC s. -25 ... +85 ºC

At nominal pressure1)

TC = Temperature coeﬀicient

Huba Control Type 515 | Technical data subject to change | Edition 09/2020

1

Order code selection table in bar
Pressure mode

Pressure range 1)

Sealing material 2)
AdjustmentOutput / power supply
Electrical connection

Pressure connection 3)

VersionPressure range variation(optional)

9899999999Sealing material 2)
Adjustment
Output / power supply
Electrical connection

Pressure connection 3)

VersionPressure range variation(optional)
1)

40
1
1
1
1
13
3
3
3
4
4
4
4
550
12
4
5
70
1
2
30
1
2
3
4
5

5

6

7

8

90

1
62
3
40

14
98
3
7
56

1
2
5

Indicate W and state range on order (e.g.: W0... + 8bar/OUT1...9V)

W2

3

4

569899999999AB
B
B
B
CCC
DDDDEE014571230
1234589102
22,52,52,52,52,5

6
60
12
60

1
62
34

Indicate W and state range on order (e.g.: W0... + 120psi/OUT1...9V)

Pressure and flow

7515. X X X X X X X X X X

Relative
Absolute-30 ...
0"hg
0 ...
15 psi0 ...
30 psi0 ...
60 psi0 ...
100 psi0 ...
200 psi0 ...
300 psi0 ...
500 psi0 ...
750 psi
0 ... 1000 psi
0 ... 2000 psi0 ... 3000 psi
0 ... 5000 psi
FPM spec. seal only
0 ... 7500 psi
FPM spec. seal only
▲ Full scale signal at these pressures
FPM
Fluoro elastomer
EPDM
Ethylene propylene
NBR
Butadiene Acylonitrile
FPM spec.
Fluoro elastomer spec.
Factory
0 ... 5 V
8.0 ... 33 VDC
1 ... 6 V
8.0 ... 33 VDC
0 ... 10 V
11.4 ... 33 VDC
4 ... 20 mA
8.0 ... 33 VDC
ratiom. 10 ... 90%
5 VDC ±5%
Cable 1.5 m PUR
Inside thread
G ¼ with O-Ring seal
G ¼ sealed at back, DIN 3852-E
G ½ sealed at front
G ½ sealed at back and manometer (combi)
Outside thread¼ -18 NPTR ¼, EN 10226
M12x1.5 sealed at back, DIN 3852-E
M14x1.5 sealed at back, DIN 3852-E
without media stopper
Stainless steel
with media stopper (standard from 750 psi)with pressure tip orifice (from 2000 psi on)

Other pressure ranges on request

102
22,52,52,52,52,56
60
12
61

Pressure range 1)3515. X X X X X X X X X X

Relative
Absolute-1 ...
0 bar
0 ...
1 bar
0 ...
1.6 bar0 ...
2.5 bar
0 ...
4 bar
0 ...
6 bar
0 ...
10 bar0 ...
16 bar0 ...
25 bar0 ...
40 bar0 ...
60 bar0 ... 100 bar
0 ... 160 bar
0 ... 250 bar
0 ... 400 bar
FPM spec. seal only
0 ... 600 bar
FPM spec. seal only
▲ Full scale signal at these pressures
FPM
Fluoro elastomer
EPDM
Ethylene propylene
NBR
Butadiene Acylonitrile
FPM spec.
Fluoro elastomer spec.
Factory
0 ... 5 V
8.0 ... 33 VDC
1 ... 6 V
8.0 ... 33 VDC
0 ... 10 V
11.4 ... 33 VDC
4 ... 20 mA
8.0 ... 33 VDC
ratiom. 10 ... 90%
5 VDC ±5%
Cable 1.5 m PUR
Inside thread
G ¼ with O-Ring seal
G ¼ sealed at back, DIN 3852-E
G ½ sealed at front
G ½ sealed at back and manometer (combi)
Outside thread¼ -18 NPTR ¼, EN 10226
M12x1.5 sealed at back, DIN 3852-E
M14x1.5 sealed at back, DIN 3852-E
without media stopper
Stainless steel
with media stopper (standard from 40 bar)with pressure tip orifice (from 100 bar on)

Order code selection table in psi
Pressure mode22)

Other sealing material on request

014983756125W3)

Other pressure connection on request

Huba Control Type 515 | Technical data subject to change | Edition 09/2020

3/6

1

Order code selection table in MPa
Pressure mode

Pressure range 1)

Sealing material 2)
Adjustment
Output / power supply
Electrical connection

Pressure connection 3)

VersionPressure range variation(optional)

1)

4/6

2

3

4

569899999999F
GGGGGH
H
H
H
KKKKL
L

01
245701230123459102
22,52,52,52,52,5660126016234Indicate W and state range on order (e.g.: W0... + 0.8MPa/OUT1...9V)

Pressure and flow

8515. X X X X X X X X X X

Relative
Absolute-0.1 ...
0 MPa
0 ...
0.1 MPa
0 ... 0.16 MPa
0 ... 0.25 MPa
0 ...
0.4 MPa0 ...
0.6 MPa
0 ...
1 MPa
0 ...
1.6 MPa0 ...
2.5 MPa
0 ...
4 MPa
0 ...
6 MPa
0 ...
10 MPa0 ...
16 MPa0 ...
25 MPa0 ...
40 MPaFPM spec. seal only
0 ...
60 MPaFPM spec. seal only
▲ Full scale signal at these pressures
FPM
Fluoro elastomer
EPDM
Ethylene propylene
NBR
Butadiene Acylonitrile
FPM spec.
Fluoro elastomer spec.
Factory
0 ... 5 V
8.0 ... 33 VDC
1 ... 6 V
8.0 ... 33 VDC
0 ... 10 V
11.4 ... 33 VDC
4 ... 20 mA
8.0 ... 33 VDC
ratiom. 10 ... 90%
5 VDC ±5%
Cable 1.5 m PUR
Inside thread
G ¼ with O-Ring seal
G ¼ sealed at back, DIN 3852-E
G ½ sealed at front
G ½ sealed at back and manometer (combi)
Outside threa d¼ -18 NPTR ¼, EN 10226
M12x1.5 sealed at back, DIN 3852-E
M14x1.5 sealed at back, DIN 3852-E
without media stopper
Stainless steel
with media stopper (standard from 4 MPa)
with pressure tip orifice (from 10 MPa on)

Other pressure ranges on request

72)

Other sealing material on request

01498375
6125W3)

Other pressure connection on request

Huba Control Type 515 | Technical data subject to change | Edition 09/2020

Dimensions in mm / Electrical connections

2-wire(4 ... 20 mA)

3-wire

brown

brown

white
Pressure absolute
Pressure relative~ 46~ 45

Pressure absolutePressure relative~ 46~ 45

greengreen

G 1/4
Inside threadius

3=9dinenn. bmidg ra

All absolute versions are especiallymarked with an indentation

manometer connector

Pressure and flow

Huba Control Type 515 | Technical data subject to change | Edition 09/2020

5/6

www.hubacontrol.com