T 9555 EN
Media Series
Accessories for Differential Pressure and Flow Meters

Application
The accessories listed in this information sheet upgrade the devices of the Media series and customize them to the plantdepending on the specific application. The following pages contain exact descriptions of all accessories with order numbers.

SAMSON AKTIENGESELLSCHAFT · Weismüllerstraße 3 · 60314 Frankfurt am Main, Germany
Phone: +49 69 4009-0 · Fax: +49 69 4009-1507 · samson@samsongroup.com · www.samsongroup.com

Edition June 2022

Contents

Media 05

2

Media 5Media 6/6Z

Media 7

Sec.

Accessories/designation

Page

1

Valve block with shut-off valves and test connection

3

2

Pressure gauge for valve block5

3

Pipe fittings · Screw fitting/cap kits

64Compensation chambers

7

5

Mounting material for wall and pipe mounting

7

6

Limit contacts · Analyzing instruments for limit contacts

87

Range springs

9

8

Electric pressure transmitter for direct attachment

9924 V power supply unit with enclosure (IP 65)

11

10

Optional additional functions for Media 7 and SAM Connect
Gateway12

11

Sun shield for Media 5 and Media 71212Media 5 · Option module with 4 to 20 mA current output

13

13

TROVIS-VIEW configuration software

14

14

SAM TANK MANAGEMENT

15

T 9555 EN

1. Valve block with shut-off valves and test connection
Three valves are combined in the valve block for Media devices. The valve block is bolted onto the bottom of the device. It offers
the following advantages:–

Mounting of an operating pressure gauge possible.

–

Fastened using two additional mounting holes.

–

The connected measuring lines can be bypassed. This allows a zero calibration to be performed regardless of the current fillinglevel of the tank.

–

The differential pressure meter can be easily removed by shutting off the measuring lines (e.g. to replace a defective device)without disturbing the running process.

–Lead-seal holes

Fig. 1: Valve block for Media devices (left) · Valve block, mounted on a Media 5 device (right)

Table 1: Technical data and materials ∙ Order numbers
Material

Minimum
temperature(material)

Maximum
temperature withMedia device

Order no.

Enclosure

SpindleO-rings

Approval foroxygenCryogenics (free of oil andgrease for oxygen),
DIN EN ISO 23208

Brass

Brass

SiliconeYes: –40 to +60 °C–40 °C+80 °C

1400-7077

Cryogenics (free of oil andgrease for oxygen),
DIN EN ISO 23208

CrNiMo steel

CrNiMo steel

SiliconeYes: –40 to +60 °C–40 °C+80 °C

1400-7078

Tobacco industry/compatible
with paint

Brass

Brass

NBR

No

–20 °C+80 °C

1400-7726

CrNiMo steel

CrNiMo steel

SiliconeNo–25 °C+80 °C

1400-7397

Possible fields of application 1)

Food industry/Food 2)
Flammable gases

Brass

Brass

FPM

No

0 °C

+80 °C

1400-7794

Flammable gases

CrNiMo steel

CrNiMo steel

FPM

No

0 °C

+80 °C

1400-9184

Standard/cryogenics/
flammable gases

Brass

Brass

NBR

No

–30 °C+80 °C

1400-7160

Standard/cryogenics/
flammable gases

CrNiMo steel

CrNiMo steel

NBR

No

–30 °C+80 °C

1400-7161

Flammable liquids andrefrigerants

Brass

Brass

EPDM

No

–30 °C+80 °C

1400-7795

1)
2)

Refer to spec. sheet 1010-4301 for more details on these versions and their requirements
Not certified for FDA compliance

Table 2: Screw fittings, locking plugs etc. · Free of oil and grease for oxygen according to DIN EN ISO 23208
Screw fittings, locking plugs etc.

Order no.

Locking plug G ½ LH for valve block pressure gauge connection, brass

1400-7873

Locking plug G ½ LH for valve block pressure gauge connection, stainless steel

1400-7874

Adapter for Media valve block test connection, 8 mm screw fitting, stainless steel

1400-9564

M20x1.5 locking plug for valve block test connection, stainless steel

1400-9745

Two 18x2.5 NBR O-rings for dp-cell connection to valve block

1400-9477

One 13x1.5 NBR O-ring for G ½ pressure gauge connection

1400-9562

One valve block knob (1400-7077 and higher, index 7)0049-0019

T 9555 EN

3

Mounted valve block (onto a Media 7 device inthe example)

5
1

3

2

4
6

7
1
2
3
4
5
6
7

Test connection
Shut-off valve (–)
Equalizing valve
Shut-off valve (+)
Fastening holes (Ø9) for wall mounting
Combined connection G ½ for pressure gauge NG 100 and G ¼ for pressure gauge NG 63
Screw fittings for process connections

Valve block · Dimensions in mm
35

25

Ø9

117

30

8

97

31
60

26

max. 160
Fig. 2: Valve block · Attachment and dimensions

4

T 9555 EN2. Pressure gauge for valve block
Pressure gauges NG 63 and NG 100
Table 3: Technical data and materials
Versions
Safety class

S3 (EN 837-1)
Safety pressure gauge for oxygen with solid baffle wall, blowout rear panel andred pointerCleaning

Free of oil and grease, for oxygen according to DIN EN ISO 23208

Pressure ranges

0 to 2.5 through 0 to 60 bar

Operating temperature–40 to +60 °C

Degree of protection according to
DIN EN 65029

IP 54
Pressure gauge NG 100

Connection
NG 100G ½ B (rear)

NG 63

G ¼ B (rear)

Note

Observe the selection criteria forpressure gauges (concerning
safety) according to EN 837-2(see also u AB 09).

Materials
Housing

CrNi steel

Wetted parts

CrNi steel

WindowLaminated safety glass

Table 4: Order numbers for pressure gauges NG 63 and NG 100
PN 2.5

PN 4

PN 6

PN 10

PN 16

PN 25

PN 40

Pressure gauge NG 63, EN 837-1, S3

Pressure rating

8520-2035

8520-2036

8520-2037

8520-2038

8520-2039

8520-2040

8520-2041–

Pressure gauge NG 100, EN 837-1, S3

8520-2042

8520-1320

8520-1321

8520-1322

8520-1323

8520-1324

8520-1325

8520-2043

–
–

PN 60

Optional versions (on request)
Pressure gauge NG 100 with limit contacts on request

Dimensions
60

2

13

3
61

Pressure gauge NG 63, safety class S3

Ø101

SW22

34

G½ B
Ø6

Ø64

SW14
18

G¼ B

Ø5

41

20
92

Pressure gauge NG 100, safety class S3

Fig. 3: Dimensions ∙ Pressure gauges NG 63 and NG 100

T 9555 EN

53. Pipe fittings · Screw fitting/cap kits
Pipe fittings with restriction · To dampen vibrations of the process medium in the measuring line with integrated restriction (especially for gas flow measurement) · Compression-type fittings G 3/8 · To connect the differential pressure lines with outside pipe diameters 6, 8, 10 and 12 mm · Screw plugs G 3/8 · To seal the dp-cell · Vent plugs G 3/8 · To vent the dp cell · Additional screw fittings · Screw fittings for attachment of Media 05/5 to Media 6 · Special screw fittings
Table 5: Screw fittings ∙ Order numbers
Order no.
Pipe fittings, vent plugs, screw plugs etc.

Standard

Free of oil and grease foroxygenG 3/8 compression-type fitting for 12 mm pipe outside diameter, steel

1400-5842–

G 3/8 compression-type fitting for 12 mm pipe outside diameter, stainless steel

1400-5844

1400-5845

G 3/8 compression-type fitting for 10 mm pipe outside diameter, steel

1400-5846–

G 3/8 compression-type fitting for 8 mm pipe outside diameter, steel

1400-5860–

Brass vent plug including copper flat gasket

1400-5654

1400-5658

Brass screw plug including copper flat gasket

1400-5655

1400-5659

Four copper flat gaskets

1400-5660

1400-5661

Two brass G 3/8 screw plugs · Two NBR O-rings

1400-7643

1400-7644

Two 15.6x1.78 O-rings for screw plugs, NBR–

1400-9563

Two stainless steel G 3/8 screw plugs–

1400-7872

Screw fitting for attachment of Media 05/5 to Media 6, brass–

1400-7748

Screw fitting for attachment of Media 05/5 to Media 6, stainless steel–

1400-7762

Two screw fittings G ½ to G 3/8 CuZn, including O-rings (WIKA conversion kit)–

1400-7749

Special screw fitting in stainless steel, free of oil and grease for oxygen, with restriction made of brass (CW617N), restriction bore: Ø0.5 mm
G 3/8 straight male connector, CrNi steel

For pipe Ø in mm

Order no.

6

1400-9108

ErmetoErmeto8

1400-9109

Swagelok®

8

1400-9110

Ermeto10

1400-9111

Ermeto 2553

12

1400-9112

Screw fitting/cap kits for attachment to dp cell and valve block · Free of oil and grease for oxygen · Additional kits on request
Table 6: Screw fitting/cap kits · Order numbers
Attachment to dp cell

Attachment to valve block

Order no.

Two brass screw plugs · Two NBR O-rings

Two 8 mm stainless steel screw fittings

1400-8823

One brass screw plug · One NBR O-ring
One 8 mm stainless steel screw fitting (high-pressureconnection)

Two 10 mm stainless steel screw fittings

1400-8824

Pipe fittings with/without restriction

Screw plugs, screw fitting/cap kits

Screw fitting to combine Media 05/5 andMedia 6

Fig. 4: Screw fittings

6

T 9555 EN

4. Compensation chambers
Compensation chambers · To build a liquid column above the measuring unit (compulsory for steam measurement) · Can also be
used as separation chamber for gas measurement by changing the arrangement of screw fittings
Table 7: Technical data and materials
Type A

Compensation chambers 1)

Type 3994-9002

Max. perm. operating gaugepressure

40 bar

35 bar

20 bar113 bar105 bar

93 barMax. permissible temperature 2)

120 °C

250 °C

400 °C

120 °C

250 °C

400 °C

Weight0.9 kg1.25 kgBody

1.0037 (St 37-2)

1.4571

Spindle––Materials · Material numbers according to DIN EN

1)
2)

Can also be used as separation chamber for gas measurement by changing the arrangement of screw fittings
Max. permissible temperature according to DIN EN 837-1; shut-off valves and pressure gauges must be protected against a temperature increase caused by hot process media through measuring lines or siphons of appropriate length.

Table 8: Order numbers
Type

Connection

Compensation chamber, Type A
Compensation chamber, Type 3994-9002

1080-0261
3994-9002

Fig. 5: Compensation chamber
35°

35°

Ø89

35°

155

105

35°

G 3/8 compression-type
fitting for 12 mm pipe

Order no.

Ø130

110

Compensation chamber, Type A

Compensation chamber, Type 3994-9002

Fig. 6: Dimensions ∙ Compensation chambers

5. Mounting material for wall and pipe mounting
Mounting material for all devices of the Media series
Pipe clips for 2" pipe mounting or holding brackets for wall mounting
Table 9: Order numbers
Mounting material

Order no.

Mounting material for 2" pipe mounting

1400-5656

Mounting material for wall mounting

1400-5657

Fig. 7: Mounting material

T 9555 EN

7

6. Limit contacts · Analyzing instruments for limit contacts
Table 10: RoHS-compliant retrofit contact module for Media 05/5 · Order numbers
Retrofit contact module

Function

Order no.

Two inductive limit contacts, SC3,5-NO-BU, acc. to ATEX

Two universal adjustable contacts

1400-8839

Three inductive limit contacts, SC3,5-NO-BU, acc. to ATEX

Two min. and one max. contacts

1400-8840

Two inductive limit contacts, SJ3,5-SN, acc. to ATEX, SIL 3

Two universal adjustable contacts

1402-1772

One inductive limit contact, SJ2-SN, acc. to ATEX

One min. contact

1402-1773

Two inductive limit contacts, SJ2-SN, acc. to ATEX

Two min. contacts

1402-1774

Two inductive limit contacts, SJ2-SN, acc. to ATEX

One min. and one max. contacts

1402-1775

Media 5Media 05

Function
Min. contact: metal tag moves in when the reading decreases · Max. contact: metal tag moves in when the reading increases ·Metal tag inside pick-up field: OFF switching signal (0 signal of the proximity switch) · Contact open or output effectively non-conducting, high resistance (damped), power consumption ≤ 1 mA.
Metal tag outside pick-up field: ON switching signal (L signal of the proximity switch) · Contact closed or output effectively conducting, low resistance (undamped), power consumption ≥ 3 mA.
Analyzing instruments for limit contacts
System K isolating switch amplifier: the isolating switch amplifiers transfer the digital signals of the limit switch (alarm contacts).
Table 11: Technical dataType

KFD2-SR2-Ex…

KFA5-SR2-Ex…

KFA6-SR2-Ex…

Power supply

20 to 30 VDC

103.5 to 126 VAC

207 to 253 VAC

Nominal data according to DIN 19234 or NAMUR
Open-circuit voltage/short-circuitcurrent

Approx. 8 VDC/approx. 8 mA

Switching point/switching accuracy

1.2 to 2.1 mA/approx. 0.2 mA

Input pulse length/input pulseinterval≥20 ms/≥20 ms

Line monitor

Rupture I ≤0.1 mA
10.5 V

10.6 V

13 mA

19 mA

Power P0

34 mW

51 mW

Fig. 8: System K isolating
switch amplifier

20

115

Table 12: Order numbers
Type

Implementation

Order no.

KFA6-SR2-Ex1.W

1-channel

8812-0100

KFA6-SR2-Ex2.W

2-channel

8812-0103

KFA5-SR2-Ex1.W

1-channel

8812-0099

KFA5-SR2-Ex2.W

2-channel

8812-0102

KFD2-SR2-Ex1.W

1-channel

8812-0098

KFD2-SR2-Ex2.W

2-channel

8812-0101

In terminal housing

On request

In alarm system

On request

111

Voltage U0
Current I0

118

Maximum values according to certificate of conformity PTB 00 ATEX 2081

Fig. 9: Dimensions ∙ System K isolating switch amplifier

8

T 9555 EN

7. Range springs
Table 13: Range springs for Media 05/5 · Free of oil and grease for oxygen
dp cell

Diaphragm

1

Diaphragm 0.4/60

1Diaphragm 0.4/60

2Diaphragm 0.6/60

Measuring span

Order no.

0 to 60 mbar

1400-7822

0 to 100 mbar

1400-7823

0 to 160 mbar

1400-7824

0 to 250 mbar

1400-7825

0 to 400 mbar

1400-7826

0 to 600 mbar

1400-7827

0 to 1000 mbar

1400-7828

0 to 1600 mbar

1400-7829

0 to 2500 mbar

1400-7830

0 to 3600 mbar

1400-7831

Tool to mount range springs and diaphragm

1180-9907

Note
The range springs of Media 6 can only be changed by the manufacturer.
Table 14: Spare diaphragms for Media 05/5/6 · Free of oil and grease for oxygen
Diaphragm

Measuring span

Order no.

ECO diaphragm 0.4/60

0 to 400 mbar

1402-0664

ECO diaphragm 0.4/60

600 to 1600 mbar

1402-0665

ECO diaphragm 0.6/60

2500 to 3600 mbar

1402-0666

Tool to mount range springs and diaphragmMedia ...
05/5/61180-9907

Note
Read the mounting and operating instructions before replacing the range springs and diaphragm.
8. Electric pressure transmitter for direct attachment
Table 15: Technical data and materials · Free of oil and grease for oxygen (according to DIN EN ISO 23208)
Pressure transmitters
Cleaning

No explosion protection

With explosion protection 1)

Oxygen according to DIN EN ISO 23208, freeof oil and grease

Cryogenic gases according to
DIN EN ISO 23208, free of oil and grease

Supply voltage

10 to 36 V DC

10 to 28 V DC

Output signal

Two-wire system (4 to 20 mA)

Two-wire system (4 to 20 mA)

Electrical connection

4-pin M16 connector

4-pin M16 connector

0 to 4 through 0 to 60 bar

0 to 4 through 0 to 60 bar

Measuring accuracy±1 % of URL 2)±0.25 % of URL 2)

Permissible ambient temperature–30 to +60 °C–20 to +60 °C

Process medium temperature–30 to +100 °C–40 to +125 °C

IP 65

IP 65

G ½ thread

G ½ thread–

ATEX: II 1G Ex ia IIC T4 + 1D
IECEx: Ex ia IIC T4 Ga + Da

Housing1.43051.4435GasketNBR

FKM

Pressure ranges

Degree of protection according to
EN 60529
Process fluid connection
Type of protection
Materials · Material numbers according to DIN EN

1)
2)

The technical data for the explosion-protected devices may be restricted by the limits
Upper reference limit (URL) of entire measuring range

T 9555 EN

9~135*

75

75

Ø27

Ø27

20

SW 24

20

SW 24

Screw fitting
for mounting

Ø6

G½B

G½B

No explosion protection

Ø6

With explosion protection*: Overall height after mounting

Mounted on low-pressure connection of dp cellFig. 10: Dimensions ∙ Pressure transmitter

Table 16: Order numbers
Order no.
Pressure measuring
range

No explosion
protection

With explosion
protection

0 to 4 bar

8523-0295

100090158

0 to 6 bar

8523-0296

100090159

0 to 10 bar

8523-0297

100090160

0 to 16 bar

8523-0298

100090161

0 to 25 bar

8523-0299

100090162

0 to 40 bar

8523-0300

100090163

0 to 60 bar

8523-1006

100090164

Screw fitting for mounting a pressure gauge with radialconnection and pressure sensor · Free of oil and grease foroxygen–Coupling sleeve G ½ (stainless steel)–

8x3 O-ring, NBR–

G 3/8 x G ½ LH screw fitting (stainless steel)Screw fitting for
mounting

Pressure transmitter screwed
into the dp cell of a Media 6
deviceFig. 11: Screw fitting for mounting

Table 17: Screw fitting for mounting · Order numbers
Screw fitting for mounting

Order no.

Coupling sleeve, O-ring and screw fitting

1400-7642

10

T 9555 EN

9. 24 V power supply unit with enclosure (IP 65)
For power supply of Media 7 devices; hardware package consisting of:–

24 V power supply unit for rail mounting–

Compact electrical distribution board with degree of protection IP 65–

Two cable glands

Table 18: Technical data · 24 V power supply unit withenclosureOutput voltage:
24 V DC

24 V power supply unit
Input voltage

90 to 264 V AC

Output voltage

24 V DC

Max. output current

1.25 A

Power

30 W

Insulation (voltage)

3 kV

Permissible ambient temperature–25 to +71 °C

Compact electrical distribution board
Enclosure material

PC with glass fiber reinforcement

Color

RAL 7035

Degree of protection

IP 65

Permissible ambient temperature–35 to +80 °C

Max. relative humidity

95 % (at +25 °C)

Installation

DIN terminal rail

WeightApprox. 0.8 kg

Fig. 12: 24 V power supply unit

~1701402-1806

150

Order no. (power supply unit +compact electrical distribution
board)

Input voltage:
90 to 264 V AC

~96

100

Fig. 13: Dimensions · Compact electrical distribution board with integrated power supply unit

T 9555 EN

1110. Optional additional functions for Media 5 and Media 7 Differential Pressure Meters as well as SAM Connect Gateway
Media 7The modular design of the Media 7 Differential Pressure Meter
allows it to be tailored to meet specific requirements. Additional functions are available through the use of option modules.
SAM Connect Gateway
Upon delivery of the SAM Connect Gateway, at least one option module is installed. Further option modules can be retrofitted.
Option modules (for subsequent installation):
AI: Analog input
The AI: Analog input option module accepts 4 to 20 mA
signals from filling level or pressure sensors of external
equipment with their own power supply. This module
works passively and has galvanically isolated inputs.
Î Order no. 1402-1809 (without explosion protection)–

AIA: Analog input active
The AIA: Analog input active option module accepts 4 to
20 mA signals from filling level or pressure sensors of external equipment. This module works actively and has a
12 V output to power external equipment that do not havetheir own power supply.
Î Order no. 1402-1810 (without explosion protection)–

Fig. 14: Option module for Media 7 and SAM Connect Gateway

AO: Analog output (Media 7 only)
The AO: Analog output option module issues an internalmeasuring signal (4 to 20 mA) representing the tank pressure or, depending on the operating mode, the filling level
or differential pressure. The analog output parameters canbe configured.
Î Order no. 1402-1808 (without explosion protection)
Î Order no. 1402-2086 (Ex IIB)–

11. Sun shield
Suitable to protect the differential pressure meter from direct
sunlight and in mounting situations where direct sunlightmakes it difficult or impossible to read the display.
Table 19: Ordering data
Sun shield

Order no.

For Media 5 and Media 7

100112667

Fig. 15: Sun shield for Media 5 and Media 7

12

T 9555 EN12. Media 5 · Option module with 4 to 20 mA current output
The option module with 4 to 20 mA current output can be
added to the standard version of the Media 5. It is installed in
the indicating unit and can either be ready installed in the device on delivery or retrofitted later.

Note
Details on option module with 4 to 20 mA current output
u EB 9519-1.

The current output module upgrades the analog reading of thedevice by issuing the current signal which can be used as areading or for further processing.
The angle of the pointer axis of the Media 5 is measured by
the magnetoresistive measuring system and converted into anelectric signal. To operate the option module with 4 to 20 mA
current output, a transmitter supply voltage of UB = 12 to 36 V
(DC) is required for the 4 to 20 mA measuring circuit.

Option module, installed into Media 5

Fig. 16: Option module with 4 to 20 mA current output

Table 20: Technical data
4 to 20 mA current output

Module 100049064 with explosion protection 1)

Version
Supply voltage UB
Output signal

Module 100033844 without explosion protection

Magnetoresistive measuring system
12 to 28 V DC

12 to 36 V DC

4 to 20 mA, two-wire system according to DIN 66258

Perm. load RB in Ω

RB = (UB – 12 V)/0.020 A|(R ≤600 Ω at 24 V and 20 mA)

Power consumption

0.252 mW for 12 V DC and 21 mA

Settings
Characteristic

Zero calibration|Span calibration|Characteristic selection|Test function
Output and reading linear or square root extraction depending on installed flow characteristic
Characteristic set at the factory

Deviation from terminal-based
linearity<±0.2 %, related to 270° measuring span

Sensitivity<±0.05 %, related to 270° measuring span

Effect of ambient temperature in
the range from –40 to +80 °C
Type of protection<0.1 %/10 K for zero and span
ATEX: II 2 G Ex ia IIC T4 Gb
IECEx: Ex ia IIC T4 Gb–

Conformity
1)

The technical data for the explosion-protected devices may be restricted by the limits specified in the test certificates.

T 9555 EN

13

13. TROVIS-VIEW configuration software
TROVIS-VIEW provides a uniform user interface that allows
users to configure and parameterize various SAMSON
devices using device-specific database modules. The TROVISVIEW software enables the user to easily configure theconnected device as well as view process parameters online.

TROVIS-VIEW for Media 7 and SAM Connect Gateway

Note
TROVIS-VIEW can be downloaded free of charge from our
website at www.samsongroup.com > Service & Support >Downloads > TROVIS-VIEW.

Media 6 devices need to be connected to the serial interfaceof the computer using the appropriate connecting cable. Depending on the computer used, a USB to RS-232 adapter may
additionally be required.

Accessories

For this purpose, the device has a SAMSON serial interface(SSP) to allow the USB port of a computer to be connected toit using an adapter cable.
TROVIS-VIEW for Media 6

Memory pen (optional): data can be uploaded from the RS232 port of an installed Media 6 device to the memory pen.

Use withMedia 7

Media 6

SAM Connect Gateway

Order no.

Isolated USB interface adapter••

1400-9740

Modular adapter 1)•

1400-7698•

1400-7699•

1400-9753•

8812-2001

Connecting cable 1)

Memory pen 1)

USB to RS-232 adapter

1)

14

Also included in the hardware package (order no. 1400-9998)

T 9555 EN14. SAM TANK MANAGEMENTSAM TANK MANAGEMENT is a web-based application specifically developed for monitoring the filling levels of liquids,
gases and vapors stored in stationary or truck-mounted pressure vessels.
Any smart mobile device can be used to access SAM TANKMANAGEMENT.
The Media 7 Differential Pressure Meter and the SAM Connect
gateway are suitable for wireless connection to SAM TANKMANAGEMENT. This ensures safe data exchange all acrossthe world, polling of states, as well as monitoring and operation of the Media devices.
Additionally, the Media 7 Differential Pressure Meter and the
SAM Connect gateway function as a hub, which can connect
up to four Media devices to SAM TANK MANAGEMENT and
as a result, monitor several tank farms at the same time.

Liquid level measurement

Wireless connection to SAM TANK MANAGEMENT using mobile phone networks

Media 7 with GSM module

Remote parameter setting
and unrestricted access

Data exchange all across the world_

SAM TANKMANAGEMENT

WWW

Ar, N2,
O2, CO2+

Fig. 17: Sample application of SAM TANK MANAGEMENT

T 9555 EN

15

T 9555 EN

2023-08-08 · English

Specifications subject to change without notice