XFL212R
Compression Miniature Load Cell
SPECIFICATIONSFEATURES

Load Washer Model
Flat sensor
High Stiffness
Integrated Spherical Load Button
Other designs available onrequest

APPLICATIONSStrain table measurement
Micro component assembly tools
Mechanical switches control
Laboratory
Robotic

SENSOR SOLUTIONS /// XFL212R_en

Ranges from 0-5 to 0-500N [0-1.12 to 112 lbf]
Extremely Flat 3,5 mm [0.14”]
Diameter Only 12 mm [0.49”]
Integrated Spherical Load Button

The XFL212R series is an extraordinarily thin miniature load cellwith a temperature compensation module integrated into theoutput cable. This design allows TE CONNECTIVITY tomanufacture extremely small sensors without sacrificing thermal
zero and sensitivity performance.
The XFL212R measures strain during compression in
static and dynamic applications. Unlike sensors with flat force
application surfaces, the XFL212R incorporates a spherical loadbutton, which assures more precise loading and in return moreaccurate measurements.
The sensing element is fitted with a fully temperaturecompensated Wheatstone bridge equipped with high stability
micro-machined silicon strain gages. The use of silicon strain
gages optimizes its performance at low ranges and frequencies.
The sensor is available in aluminum alloy or stainless steel,depending on the full scale range and can withstandconsiderable overloads.
On request, Instruction documents can be provided toease the selection and use of our sensors and provide helpfultips.”

04/2019

Page 1

XFL212R

Miniature Load CellPERFORMANCE SPECIFICATIONS (typical values at temperature 23°C)
Ranges (FS)

5N

10 N

20 N

50 N

100 N

200 N

500 N

Ranges (lbf)

1.12

2.25

4.5

11.2

22.5

45112

Material

Stainless Steel

Aluminium

Stiffness (N/m)

8.7E+05 1.9E+06 4.2E+06 1.1E+07 2.0E+07 4.0E+07 5.6E+07

Stiffness (lbf/ft)

6.0E+04 1.3E+05 2.9E+05 7.5E+05 1.4E+06 2.7E+06 3.8E+06

Sensitivity (FSO)

120 mV

150 mV

Specifications
Power supply

10Vdc

Overrange Without Damage

2 x FS

Overrange Without Destruction

3 x FS

Offset<±10 mV

Input Impedance

1500 ohms

Output Impedance

500 ohms

Linearity< ±1% de l'EM

Hysteresis< ±1% de l'EM

Operating Temperature Range (OTR)-40°C to +120°C (-40 to 250°F)

Compensated Temperature Range (CTR)

0°C to +60°C (32 to 140°F)

Thermal Zero Shift in CTR< 2% de l'EM/50°C

Thermal Sensitivity Shift in CTR< 2%/50°C

Insulation> 100 Mohms

Protection Index

IP50Notes
1. Signal goes negative in compression with standard wiring configuration
2. Electrical Termination: Shielded cable with 4 wires (AWG36/28), standard length 2 m [6.5 ft] with strain relief spring
3. Material: Body in stainless steel or aluminum alloy
4. Protection Index: IP50
5. CE conformance according to EN 61010-1, EN 50081-1, EN 50082-1

SENSOR SOLUTIONS /// XFL212R_en

04/2019

Page 2

XFL212R

Miniature Load CellDIMENSIONS & WIRING SCHEMATIC (METRIC)
Wiring Schematic

Dimensions in mm [inch]

SENSOR SOLUTIONS /// XFL212R_en04/2019

Page 3

XFL212R

Miniature Load CellOPTIONSZ0

CTR -40°C to +20°C (-40°F to 68°F)

Z04

CTR -40°C to +90°C (-40°F to 194°F)

Z1

CTR -20°C to +40°C (-4°F to 104°F)

Z3

CTR +20°C to +80°C (68°F to 176°F)

Z35

CTR +20°C to +120°C (68°F to 248°F)

L5M,L10M,L15M

Special total cable length (ex: "L05M= 5m length).(Standard total cable length = 2m)

M0.25M , M0.5M

Special distance between sensor and module(ex:M0.5M=0.5m) .(Standard distance =1m)

ORDERING INFO

XFL212R- 10KN - /L5M
Options (Z0, L3M,..)
Range in Newton
ModelTE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred to herein
might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard
Terms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuseof the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies

SENSOR SOLUTIONS /// XFL212R_en

All Rights Reserved.

04/2019

Page 4