AVAILABLE
with Signature
Calibration

SEE PG. IN-5High Range
Wet/Wet Differential
Pressure Transducers
Models Z and A-5

AMPLIFIED OUTPUT AVAILABLE

1-888-282-9891

2000 to 10,000 psid
ACCURACY TO 0.25%
The Z and A-5 High Range Wet/Wet transducers are engineered to measure differential pressures as great as 10k psid and achieve accuracies of 0.25%-0.5% full scale. Each series isbi-directional and accepts fluid in both ports. 17-4 PH stainless steel ensures durability of thesebonded foil strain gage units under harsh industrial conditions. Typical applications include flow
measurement, depth sensing, pressure equalization, and hydraulic testing.

Sensotec Sensors

Model Z

Order Code: BD142PERFORMANCE

Pressure Ranges..........................
Accuracy.......................................
Non-Linearity (max)......................
Hysteresis (max) ..........................
Non-Repeatability (max)...............
Output (standard) .........................
Resolution ....................................

2000 to 10,000 psid
+ / – 0.25% F.S.
+ / – 0.15% F.S.
+ / – 0.10% F.S.
+ / – 0.05% F.S.
2mV/VInfinite

2000 to 10,000 psid
+ / – 0.5% F.S.
+ / – 0.25% F.S.
+ / – 0.13% F.S.
+ / – 0.07% F.S.
2mV/V
Infinite

ENVIRONMENTAL

Temperature, Operating ...............
Temperature, Compensated ........
Temperature Effect– Zero (max).............................
– Span (max)............................

– 65° F to 250° F
60° F to 160° F– 65° F to 250° F
60° F to 160° F

0.5% F.S/100° F
0.5% Rdg./100° F

0.75% F.S/100° F
1% Rdg./100° F

Bonded foil
10VDC
350 ohm#2 (See Pg. AP-8)
PTIH-10-6P or equiv.
(Hermetic stainless)
PT06A-10-6S or equiv.

Bonded foil
10VDC
350 ohm#2 (See Pg. AP-8)
PTIH-10-6P or equiv.
PT06A-10-6S or equiv.

Gas, Liquid
2000 psi

Gas, Liquid
2000 psi

100% over capacity
50% over capacity
1/4-18NPT female
17-4 PH Stainless17-4 PH Stainless

100% over capacity
50% over capacity
1/4-18NPT female
17-4 PH Stainless17-4 PH Stainless

ELECTRICAL

Strain Gage Type .........................
Excitation (calibration) ..................
Bridge Resistance ........................
Wiring Code (std) .........................
Electrical Termination (std)Mating Connector (not incl.) .........

MECHANICAL

www.honeywell.com/sensing

Model A-5

Order Code: BD141Media............................................
Maximum Line Pressure ..............
Maximum Overload - safe ............
(either side)2000 to 3000 psid.....................
5000 to 10,000 psid..................
Pressure Port ...............................
Wetted Parts Material...................
Case Material ...............................

NOTES *Unless otherwise specified on order, amplified units with 4-20mA output will provide 4mA at 0psid and 20mA at positive full scale and the unit will not operate in the negative direction. Anavailable alternative is to specify 4mA at negative full scale and 20mA at positive full scale.
Intrinsically safe amp (option 2n or 2N, see page AP-6) adds 2" to amplifier housing.

Dimensions
Models Z (Order Code BD141)
Models A-5 (Order Code BD142)
Available Ranges
2,000; 3,000; 5,000; 7,500; 10,000 psid

Options (See Appendix)
Temperature compensated 1b, 1c, 1d, 1e, 1f; Internal amps 2b, 2c, 2n or 2N intrinsically safe amp (see pageAP-6), 2j, Amp enhancements 3b, 3d; Electrical termination 6e, 6f, 6g, 6h, Int. shunt cal 8a, 25b, Signature
calibration 53e.
Premium Options: 2q, 3a, 5c, 6b, 6c, 6i, 6j
Accessories: Mating connectors and connector/cable assemblies, Pressure port adapters
DP-8