MODEL 4610A ACCELEROMETERSPECIFICATIONSHigh Performance DC Response
Micro-g Resolution, Low Noise
Advanced Temp Compensation
5,000g Over-Range Protection

The Model 4610A is an ultra-low-noise
accelerometer designed for both static and dynamicmeasurements. The accelerometer offers integral
temperature compensation with dynamic range from ±2 to±100g. The model 4610A incorporates a gas damped
MEMS element with mechanical overload stops for highg shock protection. The accelerometer has an operating
temperature range of -55°C to +125°C.
For a triaxial version, TE Connectivity also offers
the model 4630A accelerometer.

APPLICATIONS
Transportation Testing
Vibration & Shock Monitoring
Road Vehicle Testing
Low Frequency Applications
Modal Analyses
Structural Monitoring

SENSOR SOLUTIONS /// Model 4610A Rev CFEATURES

±2g to ±100g Dynamic Range
5,000g Shock Protection
Signal Conditioned Output
8 to 30Vdc Excitation Voltage
Gas Damping
Integral Strain Relief
Temperature Compensated

03/2018

Page 1

MODEL 4610A ACCELEROMETERPERFORMANCE SPECIFICATIONS
All values are typical at +24°C, 80Hz and 12Vdc excitation unless otherwise stated. TE Connectivity reserves the right to update andchange these specifications without notice.
Parameters
DYNAMIC
Range (g)Sensitivity (mV/g)
Frequency Response (Hz)Frequency Response (Hz)Natural Frequency (Hz)Non-Linearity (%FSO)
Transverse Sensitivity (%)
Damping Ratio
Shock Limit (g)Residual Noise (µV RMS)
Residual Noise (µg/√Hz RMS)

±210000-2000-400
700±1.0<30.72000252

ELECTRICAL
Zero Acceleration Output (mV)
Excitation Voltage (Vdc)
Excitation Current (mA)
Bias Voltage (Vdc)
Full Scale Output Voltage (Vdc)
Output Resistance (Ω)
Insulation Resistance (MΩ)
Turn On Time (msec)
Ground Isolation±508 to 30<122.5±2<100>100<100
Isolated from Mounting Surface

Differential

ENVIRONMENTAL
Thermal Zero Shift (%FSO/°C)
Thermal Sensitivity Shift (%/°C)
Operating Temperature (°C)
Compensated Temperature (°C)
Humidity±0.010±0.014-55 to +125
-40 to +100
Epoxy Sealed, IP65

-40 to +100°C-40 to +100°C

PHYSICAL
Case Material
Cable
Weight (grams)
Mounting
Mounting TorqueAnodized Aluminum
4x #30 AWG Conductors PFA Insulated Leads, Braided Shield, TPE Jacket
7 (cable not included)
2x #4 or M3 Screws
6 lb-in (0.7 N-m)

Notes
±54000-300
0-500
800
±1.0
<30.7200020
3±10200
0-400
0-600
1000
±1.0
<30.75000
236±201000-600
0-1000
1500
±1.0
<30.7
5000
3113±50400-1000
0-15004000±1.0<30.7500026
21±10020
0-15000-2000
6000
±1.0
<30.7
5000
32
41±10%±5%±1dB<1 Typical
PassbandSpectral@100Vdc

Calibration supplied:

CS-FREQ-0100

NIST Traceable Amplitude Calibration from 20Hz to ±5% Frequency Response Limit

Supplied accessories:

AC-A02285

2x #4-40 (7/16 length) Socket Head Cap Screw and Washer

Optional accessories:

AC-D02669
121

Triaxial Mounting Block
3-Channel Precision Low Noise DC Amplifier

The information in this sheet has been carefully reviewed and is believed to be accurate; however, no responsibility is assumed for inaccuracies. Furthermore, thisinformation does not convey to the purchaser of such devices any license under the patent rights to the manufacturer. TE Connectivity reserves the right to makechanges without further notice to any product herein. TE Connectivity makes no warranty, representation or guarantee regarding the suitability of its product for any
particular purpose, nor does TE Connectivity assume any liability arising out of the application or use of any product or circuit and specifically disclaims any and allliability, including without limitation consequential or incidental damages. Typical parameters can and do vary in different applications. All operating parameters must
be validated for each customer application by customer’s technical experts. TE Connectivity does not convey any license under its patent rights nor the rights ofothers.

SENSOR SOLUTIONS /// Model 4610A Rev C

03/2018

Page 2

MODEL 4610A ACCELEROMETERDIMENSIONSSCHEMATICSENSOR SOLUTIONS /// Model 4610A Rev C

03/2018

Page 3

MODEL 4610A ACCELEROMETERTRIAXIAL MOUNTING BLOCK (PN AC-D02669)

SENSOR SOLUTIONS /// Model 4610A Rev C

03/2018

Page 4

MODEL 4610A ACCELEROMETER

ORDERING INFORMATION
4610A

GGG

ZZZRange002=2g005=5g010=10g020=20g050=50g
100=100g
Cable length
060=60 inches
120=120 inches
240=240 inches
360=360 inches
480=480 inches
600=600 inches
197=197 inches, 5 meters394-394 inches, 10 meters
Example; 4610A-002-060
Model 4610A, 2g range, 60inch (5ft) cable length

MEAS France SAS and Measurement Specialties (China) , Inc., are TE Connectivity companies.
TE Connectivity, TE, TE connectivity (logo) are trademarks. All other logos, products and/or company names referred to herein might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes nowarranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard Terms
and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuse of theproduct. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2017TE Connectivity Ltd. All Rights Reserved.

SENSOR SOLUTIONS /// Model 4610A Rev C

03/2018

Page 5