Configurable Differential
Pressure Transducers
FP2000 Series
2-week delivery
mV/V, 0-5, 0-10 VDC, OR 4-20 mA

1-888-282-9891

WET WET & WET DRY DIFFERENTIAL

The FP2000 Series is a manufacturing and delivery system which allows thecustomer to select the configuration which best fits the needs of the application. Choose
from two accuracies, four outputs, six pressure ports, five electrical terminations and twelvepressure ranges. The FP is available with wet/wet or wet/dry reference and, best of all, they
deliver in 2 weeks or less.

Wet/wet differential
FDW

Sensotec Sensors

Order CodePERFORMANCE

Positive Pressure Ranges
Accuracy (BFSL)
Output (selectable)
Resolution

See chart next page
0.1% or 0.25% F.S.
mV/V, 0-5 VDC, 0-10 VDC, or 4-20 mA (2 wire)
Infinite

ENVIRONMENTAL

Temperature, Operating
Temperature, CompensatedTemperature Error Band*
0.1% Accuracy
0.25% Accuracy-40° F to 240° F
40° F to 140° F

ELECTRICAL

www.honeywell.com/sensing

MECHANICAL±0.5% F.S.
±1% F.S.

Excitation (calibration)
Amplified (4-20mA, 0-5 VDC)
Amplified (0-10 VDC)
Unamplified (mV/V)

9 - 28 VDC
15 - 28 VDC
10 VDCMedia**
Overload-Safe (+direction)
Overload-Safe (-direction)
Wetted Parts Material
Line Pressure

Gas, Liquid
4x FS or 3,000 psi, whichever is less
4xFS or 250 psi, whichever is less
Ha C276 & 316L ss
500 psi* For ranges below 15psi, temperature effects may vary.
** The Wet/Wet differential pressure transducer has two separate, welded Hastelloy diaphragms. In the wet/dry unit, the wetport (high port) has all-welded stainless and Hastelloy construction. The dry port (low port) has no isolation diaphragm.

NOTE: Output for non-amplified units @ 10VDC excitation: 0.10% Accuracy = 50mV, 0.25% Accuracy = 100mV.

Dimensions
WET/WET
DIFFERENTIAL

CODE 6n
4-PIN, VENTEDMINI DIN

CODE 6a
6-PIN, VENTED
BENDIX STYLE

CODE 6q
4-CONDUCTORVENTED, INTEGRALCABLE, 5 FEET

WET/DRY
DIFFERENTIAL.3".3".3"

CODE 6r
4-CONDUCTORVENTED, INTEGRALCABLE, CONDUITFITTING, 5 FEET

DP-2

Wet/dry differential
FDD

CODE 6m4-PIN, VENTED
STD. DIN (43650)

1-888-282-9891
PressurePortOptions

CODE 5a1/4-18 NPT Female

CODE 5b
1/4-18 NPT Male

CODE 5c
7/16-20 UNF Female

CODE 5d
7/16-20 UNF Male

CODE 5f
G 1/4 B Female

CODE 5g
G 1/4 B Male

www.sensotec.com/FP2000.htm

HOW TO ORDER

It’s easy to order exactly what you need. Simply make one selection in each of the required categories, choose
adders and accessories only if you want them. The result is a custom transducer with quick delivery!
TypeAccuracy

Range

Output

PortConnector

FDW1AT2u

5b6aThis example is for a wet/wet differential pressure unit, 0.10% accuracy, 5 psi range, mV/V output,
1/4-18 NPT Male pressure port, and a Bendix electrical connection.

TRANSDUCER TYPE
 Wet/wet (FDW) Wet/dry (FDD)

AVAILABLE OUTPUTS mV/V (2u)
 5 VDC (2d)
 10 VDC (2g)
 4-20 mA (2p)
 4-20 mA, CE (2y)
 4-20 mA, Intrinsically Safe (2n or 2N;
limited to ranges </= 5000 psi.) See
page AP-6.

ACCURACY 0.10% (1) 0.25% (2)POSITIVE PRESSURE RANGES 10" H2O (WA) 150 psi (CJ)
 1 psi (AP) 250 psi (CN) 5 psi (AT) 500 psi (CR) 15 psi (BJ)
 750 psi (CT) 30 psi (BM) 1000 psi (CV) 50 psi (BN) 100 psi (BR)

ADDERSPRESSURE PORTS
 1/4-18 NPT F (5a)
 7/16-20 UNF M (5d)
 1/4-18 NPT M (5b)
 G 1/4 B F (5f)
 7/16-20 UNF F (5c)
 G 1/4 B M (5g)Note: The dry port on the FDD is 7/16 x 20 UN,
and is not selectable.

WIRING CODES, no shunt cal.

Selecting an Adder will automatically
update the output code.

MatingConnectors

mV/V2u0-5 VDC
2d0-10 VDC2g

AA161 Mini DIN

#37

#38

#38

#54

#53

AA111 Bendix

#2

#50

#50

#49

#23

6q Cable

#1

#52

#52

#51

#22

6r Conduit Fitting

#1

#52

#52

#51

#22

AA157 Std. DIN

#37

#38

#38

#54

#53

Output Extended temperature range (1y) Buffered shunt cal (3d) CE rating (9e) Intrinsically safe, 2 wire (9d) see page AP-6
 CE and Intrinsically safe (9f) see page AP-6
 Potentiometers (14c)

ACCESSORIESMating connectors with 15' of cable for Bendix connector (6a)
 mV/V 4-20 mA 0-5 / 0-10 VDCMating Connectors Only
 Mini DIN (40050)
 Bendix Standard DIN (43650)

without shuntwith shunt (3d)

AA113
AA116
AA117

AA513
AA516
AA517

AA161
AA111
AA157

2 WEEK DELIVERYELECTRICAL CONNECTIONS Bendix PTIH-106P (6a)
 MINI DIN, 40050 (6n)
 5´ Integral cable, polyurethane (6q)
 Standard DIN, 43650 (6m)
 Conduit fitting (6r)

DIFFERENTIAL

ExampleOrder Code:

4-20 mA
2p
2n or 2NWIRING CODES, with shunt cal. option (3d)MatingConnectors

mV/V
2u

0-5 VDC2e0-10 VDC
2f

AA161 Mini DIN

N/A

#56

#56

#55

#64

AA111 Bendix

#57

#60

#60

#58

#59

6q Cable

N/A

#63

#63

#61

#62

6r Conduit Fitting

N/A

#63

#63

#61

#62

AA157 Std. DIN

N/A

#56

#56

#55

#64Output

4-20 mA
2n or 2N
2yDP-3