FN3000
Load Cell Tension and Compression
SPECIFICATIONS




FEATURES

Static and dynamic applications
Linearity 0.1% F.S.
Integrated Amplifier optional
IP65 optional

APPLICATIONS

Process control equipment
Weighing calibration tool
Fatigue tests benches
Hydraulic press regulation
Laboratory and Research

SENSOR SOLUTIONS /// FN3000_en

Heavy duty Pan-cake load cellStandard ranges 10 kN to 1000 kN[2 klbf to 200 klbf]
Very high stability
Aluminum or Stainless steel bodyHigh IP protection available
High Level Output Model with Integrated Amplifier

The FN3000 measures tension and compression in standard
ranges from 0-10 kN to 0-1000 kN. The mechanical design andgauge placement minimizes transverse effects. Depending on
the range, the FN3000 is constructed in aluminium alloy orstainless steel and is available with numerous options. It is
suitable for test bench applications and used in many hostileenvironments and can be customized for increased protection.
With a long standing experience as a designer andmanufacturer of sensors, TE CONNECTIVITY often works withcustomers to design or customize sensors for specific uses andtesting environments.
To meet your needs we also offer extensive turnkey
systems. The matched components (sensor, power, amplifierand digital display) are formatted, calibrated and ready for
immediate use.
On request, Instruction documents can be provided toease the selection and use of our sensors and provide helpful
tips.

05/2018

Page 1FN3000

Load Cell Tension and Compression

STANDARD RANGES
Ranges (FS) (N)

10k

25k

50k

100k

200k

500k

1000k

Ranges (lbf)

2.2k

5.6k

11.2k

22.5k

45.0k

112.4k

224.8kMaterial

Aluminium

Stiffness (N/m)

4.2E+08

1.1E+09

1.4E+09

2.6E+09

Stainless Steel
3.6E+09

5.9E+098.9E+09

Stiffness (lbf/ft)

2.9E+07

7.5E+079.6E+07

1.8E+08

2.5E+08

4.1E+08

6.1E+08

PERFORMANCE SPECIFICATIONS (typical values at temperature 23°C)
VersionStandard

A1A2

Power supply

10Vdc

10Vdc à 30Vdc±12Vdc à ±18Vdc

Sensitivity (FSO)±20mV±2Vdc ±0.2V±5Vdc ±0.25V

Offset<±1mV

2.5Vdc ±0.2V

0V ±0.25V

Input Impedance / Consumption

700 ohms< 30 mA

Output Impédance

700 ohms

1000 ohms

Overrange Without Damage

1.5 x FS

Overrange Without destruction

3 x FSLinearity< ±0.1% FSHyteresis< ±0.1% FS

Operating Temperature Range (OTR)-20°C to +80°C

Compensated Temperature Range (CTR)

0°C to +60°C

Thermal Zero Shift in CTR< 0.5% FS/50°C

Thermal Sensitivity Shift in CTR< 1%/50°C

Insulation> 1000 Mohms

Protection index

IP50Notes
1. Signal goes positive in tension with standard wiring configuration. Other signal output on request
2. Electrical Termination: Connector output including mate
3. Materials: Body in stainless steel or aluminium alloy depending on F.S.; aluminum cover
4. Protection Index: IP50 (other protection levels on request)
5. Output impedance < 100Ω on request
6. CE conformance according to EN 61010-1, EN 50081-1, EN 50082-1

SENSOR SOLUTIONS /// FN3000_en05/2018

Page 2

FN3000

Load Cell Tension and Compression

DIMENSIONS & WIRING SCHEMATIC (IN METRIC AND IMPERIAL)OPTEHxxx: Hemispherical Load Button

Wiring Schematic

Dimensions in mmF.S.

JRØD

10 KN

12803425 KN

12803450 KN

1280

34

100 KN

2025065200 KN

20250

65500 KN

30400

87

1000 KN

60400120

OPTFFxxx: Tension pull plate

Dimensions in mm [inch]
Ranges in N

10k

25k

50k

100k

200k

500k

1000k

A

100 [3.94]

150 [5.91]

195 [7.68]

272 [10.71]

B

30 [1.18]

40 [1.57]

60 [2.36]

80 [3.15]

C

34 [1.34]

65 [2.56]

87 [3.43]

120 [4.72]

D (Thread)

M20x1.5

M32x2

M56x2

M80x3

E

65 [2.56]

90 [3.54]

106 [4.17]

150 [5.91]

F

70 [2.76]

100 [3.94]

143 [5.63]186 [7.32]

G45°

30°

H

8x8.2 /85

12x10.4 /125

16x16.2 /169

16x24.5 /229

M10 /125

IM8 /85

22.5°
M16 /169

M24 /229

Screw-down (m.kg)

2.2

2.5

2.5

551550Screw-down in lbf/ft

15.9

18.1

18.1

36.2

36.2

108.5

361.7

SENSOR SOLUTIONS /// FN3000_en05/2018

Page 3

FN3000

Load Cell Tension and Compression

OPTIONSZ04

CTR -40°C to +90°C (OTR -40°C to +90°C)

Z35

CTR +20°C to +120°C (OTR -20°C to +120°C)

Z36

CTR +20 to+150°C (OTR -20°C to +150°C) - Not available with A1/A2 versions

Other options may be offered as custom designed sensors. Please contact your local Customer Service.+

ORDERING INFO

FN3000 - A1 - 50KN - /Z04
Options (Z35...)
Range in Newton
Amplified version (none, A1 or A2)
Model

SUPPLIED ACCESSORIESEFMX-7M : mating plug Jaeger 530-272-006 with clamp 530-371-006 for standard and Z04 option
EFMX-7H : mating plug Jaeger 530-604-006 with clamp 530-693-006 for Z35 or Z36 options

RECOMMENDED ACCESSORIESOPTEHxxx

10kN to 1000kNHemispherical load button (Contact us)

OPTFFxxx

10kN to 1000kN

Tension pull plate (Contact us)

TE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred to herein
might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makesno warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard
Terms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuseof the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies

SENSOR SOLUTIONS /// FN3000_en

All Rights Reserved.

05/2018

Page 4