Model CIP
Ultra Sanitary Process Transmitter
These CIP (Clean-In-Place) sanitary process transmitters aremeet the environmental demands found in the applications listed

manufactured in stainless steel and offer a variety of flanges,

below in potential applications.

electrical connections, and operating temperatures designed toFEATURES•

10 psig/a to 600 psig/a range•

Track temperature changes, 93 °C/minute [200 °F/minute]•

4 mA to 20 mA output•

500 ms pressure response time•

Secondary containment exceeds 1500 psi•

Accuracy to 0.1 %•Intrinsically safe available (2N option only)9•

CE approved10

A

3Complies tostandard

Model CIPPERFORMANCE SPECIFICATIONSRANGE CODES

CharacteristicMeasure

Range CodeAvailable ranges

Pressure ranges

38,1 mm [1.5 in] flange, 10 psi to 600psi; 51 mm [2 in] flange, 10 psi to 500psi; 63,5 mm [2.5 in] flange, 10 psi to400 psi; 76 mm [3 in] flange, 10 psi to250 psi

AV10 psi

BJ

15 psi

BL25 psi

600 psi with 13MHHS clamp at 22 °C[72 °F], 38,1 mm [1.5 in]

BN50 psi

Max. flange pressure

BR100 psi

Response time(comb. mech. & elect.)

500 msCL

200 psiCN

250 psiResolution

Infinite

CQ

400 psiOrientation sensitivity

0.03 psi/GCR

500 psiCS

600 psi

ENVIRONMENTAL SPECIFICATIONS
CharacteristicMeasure

OPTION CODES

Process temperature-29 °C to 127 °C [-20 °F to 260 °F]

Sensor temperature4-1 °C to 82 °C [30 °F to 180 °F]

Range CodeTemperature effect, zero5

1 % full scale/38 °C [100 °F] (2 % fullscale/38 °C [100 °F] for 1 psi)

Many range/option combinations are available inour quick-ship and fast-track manufacture programs. Please see http://sensing.honeywell.com/
TMsensor-ship for updated listings.

Temperature effect, span5

1 % full scale/38 °C [100 °F] (2 % fullscale/38 °C [100 °F] for 1 psi)

Pressure ranges

10, 15, 25, 50, 100, 200, 250, 400, 500, 600 psi

Temperaturecompensation-1 °C to 82 °C [30 °F to 180 °F]
1i. -54 °C to 121 °C [-65 °F to 250 °F]

Internal
amplifiers

2n(2N).Intrinsically safe amp 4 mA to 20 mA,
two-wire current

2,3ELECTRICAL SPECIFICATIONS
Characteristic

MeasureSupply voltage

13 Vdc to 32 VdcInternal amplifier
Special filteringenhancements

Electrical connection (std)

1/2-14 NPT conduit exit with 1,5 m [5 ft]cable

Pressure ports

Tri-clover

Load impedance (max.)

950 ohm @ 32 Vdc

Zero balance

4 mA ± 0.16 mA

Output (std)

4 mA to 20 mA, two-wireElectricaltermination

6j. 1/2-14 NPT conduit fitting with 1,5 m  [5 ft] offour conductor PVC cable
6a. Bendix PTIH-10-6P or equivalent 6 pin
6m. DIN 436508
6n. DIN 40050

Shuntcalibration

3d. Remote buffered shunt calibration

Specialcalibration

Compound, vacuum, barometricBar

Wetteddiaphragm10c. Hastelloy C-276

Bridgeresistance

12b. 5000 ohm (foil) (max. 121 °C [250 °F])

Zero and spanadjustments

14b. Top access to pots

MECHANICAL SPECIFICATIONS
CharacteristicMeasure

Media

Steam, ethylene oxide, liquid, gas

Wetted parts material

316-L stainless, Hastelloy C-276

Fill material

NEOBEE M-20

Weight (nominal)

454 g [16 oz]

Case material

316-L stainless

Sanitary standard

3-A, #74-03

Over pressure

4X to max. flange pressure

2   Honeywell • Sensing and Control

Ultra Sanitary Process Transmitter
INTERNAL AMPLIFIERS

Mounting dimensions and characteristics

Amplifierspecifications

Intrinsically safe amp:
Option 2n(2N)***

Output signal

4 mA to 20 mA

Input power (voltage)

9 Vdc to 32 Vdc3

Input power (current)

4 mA to 20 mA

Frequency response

2000 HzPower supply rejection

60 dB

Operating temperature-29 °C to 85 °C [-20 °F to 185°F]

Reverse voltage protection

Yes

Short circuit protection

Yes

Wiring code: connector

A (+) Supply
B No connection
C No connection
D (+) Output
E Case ground
F No connection

Wiring code:
cable (std)6,7

R (+) Supply
Bl (+) Output
W Case groundFor reference only

TYPICAL SYSTEM DIAGRAM*** See Honeywell’s Web site for the most up-to-date information regarding intrinsically safe approvals, ref. #008-0547-00.*** See Honeywell’s Web site for the mostup-to-date information regarding Intrinsically safe approvals. Ref #008-0547-00.

WIRING CODES
Cable

Two-wire current, 4 mA to 20 mA

Red(+) Supply

Black(+) Output (4 mA to 20 mA)

WhiteCase ground

Flanges
CodeSizeFlange

16b*38,1 mm [1.5 in]Tri-clover

16c51 mm [2 in]Tri-clover

16d63,5 mm [2.5 in]Tri-clover

16e76 mm [3 in]Tri-clover* Does not comply to 3A standards

Special requirements (consult factory)
Have a special requirement? New case pressure, different cable lengths,
electrical connectors, or materials? Consult our factory by calling +1
614-850-5000 (800-848-6564). Customization is key to our test andmeasurement business. Special outputs, wiring codes, and calibrations
are all standard to us.

Honeywell • Sensing and Control     3Model CIP

Ultra Sanitary Process TransmitterNOTES
1.
2.
3.
4.
5.

Accuracies stated are expected for best fit straight line for all errors,including linearity, hysteresis and non-repeatability.
Higher ranges may be available for special applications. Consultfactory.
Specifications may vary with flange size and pressure range. Consult factory.
-1 °C to 54 °C [30 °F to 130 °F] for 1 psi range (only applicable for
CP100, CP101, CP200, and CP201).
Temperature profile for heated vessels:

Process temperature

Sensor temperature

127 °C [260 °F]

82 °C [180 °F]

93 °C [200 °F]

63 °C [145 °F]

66 °C [150 °F]

49 °C [120 °F]

38 °C [100 °F]

32 °C [90 °F]

24 °C [75 °F]

24 °C [75 °F]

10 °C [50 °F]

18 °C [65 °F]-18 °C [0 °F]

4 °C [40 °F]-29 °C [-20 °F]-1 °C [30 °F]

6.

Warranty. Honeywell warrants goods of its manufacture as
being free of defective materials and faulty workmanship.
Honeywell’s standard product warranty applies unless agreed
to otherwise by Honeywell in writing; please refer to yourorder acknowledgement or consult your local sales office forspecific warranty details. If warranted goods are returned toHoneywell during the period of coverage, Honeywell will repair
or replace, at its option, without charge those items it finds
defective. The foregoing is buyer’s sole remedy and is in lieu
of all warranties, expressed or implied, including those ofmerchantability and fitness for a particular purpose. In noevent shall Honeywell be liable for consequential, special, or
indirect damages.
While we provide application assistance personally, through ourliterature and the Honeywell web site, it is up to the customer todetermine the suitability of the product in the application.

G=Green; B=Blue; W=White; Bl=Black; Br=Brown; Y=Yellow;
R=Red; O=Orange.  Color specifying cable and number or letter
specifying connector.
7. No mating connector necessary with cable option.
8. For zero and span adjustment by instrumentation tech. Electricaltermination is DIN 43650 Hirschmann plug; environmental class
IP65 (NEMA 4). Conduit attachment by PG 9 thread on plug.
9. Range dependent; consult factory.
10. Termination dependent; consult factory.

Specifications may change without notice. The information we
supply is believed to be accurate and reliable as of this printing.
However, we assume no responsibility for its use.
For more information about Sensing and Control products, visit
www.honeywell.com/sensing or call +1-815-235-6847
Email inquiries to info.sc@honeywell.com

WARNINGPERSONAL INJURY• DO NOT USE these products as safety or emergency
stop devices or in any other application where failure ofthe product could result in personal injury.
Failure to comply with these instructions could result indeath or serious injury.

WARNINGMISUSE OF DOCUMENTATION• The information presented in this catalogue is forreference only. DO NOT USE this document as productinstallation information.
• Complete installation, operation and maintenanceinformation is provided in the instructions supplied witheach product.
Failure to comply with these instructions could result indeath or serious injury.

008697-1-EN IL50 GLO
May 2008
Copyright © 2008 Honeywell International Inc. All rights reserved.