FS19Force Load Cell
SPECIFICATIONS





The FS19 load cell, with ranges from 350G to3000G, uses proven Microfused sensor technology derived
from demanding aerospace and automotive applications
which employs micro-machined piezoresistive strain gages
fused with high temperature glass to a high performancestainless steel force measuring flexure. This sensing
technology eliminates age-sensitive organic epoxies usedin traditional load cell designs, providing excellent long termstability. This high sensitivity strain gage allows the load cellto produce 20 mv/V output with low strains and deflection,
thus long endurance in the smallest package. The flexureis precisely machined to produce excellent consistencyfrom part to part. Typical applications range from medical
devices to appliances and exercise equipment.

FEATURESAll stainless steel construction
Small compression load 350G to 3000G range
High sensitivity 20 mV/V
Non-linearity +/-1% FS
Flex cable for tight space

Small size
High sensitivity
High overload
Low deflection
Long life
Low cost

APPLICATIONS

Infusion Pumps
Load Sensing
Contact Sensing
Weighing
Household Appliances

SENSOR SOLUTIONS /// FS1903/2012

Page 1

FS19

Force Load CellPERFORMANCE SPECIFICATIONSAll specifications are at 25°C except where indicated.
PARAMETER
Force Range
Excitation Voltage
Sensitivities (FSO)
Non-Linearity
Hysteresis
Zero Offset
Operating Temperature
Storage Temperature
Humidity
Safe Load Limit
Insulation
Deflection
Input Resistance
Output resistances
Thermal Error – Sensitivities
Thermal Error –Zeros
Response Time
ESD
WeightWeather ProofSENSOR SOLUTIONS /// FS19MINTYP
MAX350/500/1000/2000/3000
2510142026-11-11-9+9040-4085
095
2x50
0.020
1.50K
2.15K
2.80K
1.50K
2.15K
2.80K
-0.05
+0.05
-0.5
+0.5
1±15
1.1
IP4303/2020

UNITS, NOTESGVDC
mV/V, Ratiometric to excitation voltage
%FSO
%FSO
mV/V, Ratiometric to excitation voltage
°C
°C
% RH
Rated load
Meg ohms at 250 VDCmm at rated loadΩΩ
%FSO / °C%FSO / °Cms
KV, 10 discharges qualified per IEC
gramPage 2

FS19Force Load CellDIMENSIONS (mm)SENSOR SOLUTIONS /// FS1903/2020

Page 3

FS19

Force Load Cell

ORDERING INFORMATION
FS1901

-0000

-

1000G

Model-0000-

ForceRange

FS1901-0000-0350G
0500G
1000G
2000G
3000G

TE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred to herein
might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard
Terms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuseof the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies

SENSOR SOLUTIONS /// FS19

All Rights Reserved.

03/2020

Page 4