USER’S MANUAL

6224

12 Channel  24-Bit  Half/Full-Bridge Analog Input  Ethernet

High-speed, Strain Gauge Input Module

Measurement Computing
10 Commerce Way
Norton, MA 02766*372738C-01*
372738C-01

Rev 1.2(508) 946-5100
Fax: (508) 946-9500
info@mccdaq.com

www.mccdaq.com

Warranty Information
Contact Measurement Computing by phone, fax, or e-mail in regard to warranty-related issues:
Phone: (508) 946-5100, fax: (508) 946-9500, e-mail: info@mccdaq.com

Limitation of Liability
Measurement Computing cannot be held liable for any damages resulting from the use or misuse of this product.

Copyright, Trademark, and Licensing Notice
All Measurement Computing documentation, software, and hardware are copyright with all rights reserved. No part ofthis product may be copied, reproduced or transmitted by any mechanical, photographic, electronic, or other method
without Measurement Computing’s prior written consent. IOtech product names are trademarked; other product names, asapplicable, are trademarks of their respective holders. All supplied IOtech software (including miscellaneous support
files, drivers, and sample programs) may only be used on one installation. You may make archival backup copies.

CE Notice
Many Measurement Computing products carry the CE marker indicating they comply with the safety and emissions
standards of the European Community. When applicable these products have a Declaration of Conformity stating which
specifications and operating conditions apply. You can view the Declarations of Conformity at
www.mccdaq.com/legal.aspx (CE Information page).

Warnings, Cautions, Notes, and Tips
Refer all service to qualified personnel. This caution symbol warns of possible personal injury or equipment damage
under noted conditions. Follow all safety standards of professional practice and the recommendations in this manual.
Using this equipment in ways other than described in this manual can present serious safety hazards or cause equipment
damage.
This warning symbol is used in this manual or on the equipment to warn of possible injury or death from electricalshock under noted conditions.

This ESD caution symbol urges proper handling of equipment or components sensitive to damage from electrostatic
discharge. Proper handling guidelines include the use of grounded anti-static mats and wrist straps, ESD-protective bags
and cartons, and related procedures.

This symbol indicates the message is important, but is not of a Warning or Caution category. These notes can be of great
benefit to the user, and should be read.
In this manual, the book symbol always precedes the words “Reference Note.” This type of note identifies the locationof additional information that may prove helpful. References may be made to other chapters or other documentation.
Tips provide advice that may save time during a procedure, or help to clarify an issue. Tips may include additionalreference.

Specifications and Calibration
Specifications are subject to change without notice. Significant changes will be addressed in an addendum or revision to
the manual. As applicable, the hardware is calibrated to published specifications. Periodic hardware calibration is notcovered under the warranty and must be performed by qualified personnel as specified in this manual. Improper
calibration procedures may void the warranty.

CAUTION
Using this equipment in ways other than described in this manual can cause
personal injury or equipment damage. Before setting up and using yourequipment, you should read all documentation that covers your system.
Pay special attention to Warnings and Cautions.

Note:

During software installation, Adobe® PDF versions of user manuals will automaticallyinstall onto your hard drive as a part of product support. The default location is in the
Programs group, which can be accessed from the Windows Desktop. Initial navigation
is as follows:
Start [on Desktop]

All Programs

IOtech …

Refer to the PDF documentation for information regarding hardware and software.
For detailed information regarding Encore software, refer to the Help information thatcan be accessed directly from IOtech’s Encore application following its installation.

Table of Contents
F

Quick Start
Chapter 1 – General Information
Introduction …. 1-1
Front Panel …… 1-1
Rear Panel …… 1-2
Analog Circuitry …… 1-31
Synchronization …… 1-4
Stacking Plate and Handle Options …… 1-5
Quarter-Bridge Accessories(CN-269 & CN-270) …… 1-6
RJ50 12-Pin Screw Terminal Accessory (CN-268) …… 1-7
RJ50 Cables (CA-272-01) …… 1-7

Chapter 2 – Block Diagram
Chapter 3 – Pinouts and Connections
Strain Channels, RJ50 Pinout …… 3-1
Half-Bridge and Full-Bridge Connections …… 3-2
Connecting TEDS Channels…… 3-2
Remote Sensing …… 3-2
Shunt Calibration …… 3-3
Internal Excitation Voltage Sources …… 3-3
External Excitation Voltage Sources …… 3-4
Mounting and Securing Connections …… 3-5
Where to Find Additional Information …… 3-5

Chapter 4 – Counter Input
Counter Settings …… 4-1
Encoder Settings …… 4-2

Chapter 5 – CE Compliance and Noise Considerations
Overview …… 5-1
CE Standards and Directives …… 5-1
Safety Conditions …… 5-2
Emissions/Immunity Conditions …… 5-3
CE Rules of Thumb …… 5-3
Noise Considerations …… 5-4

Chapter 6 – Software, IOtech Encore
Chapter 7 – Strain and Pressure Force
Chapter 8 – Product Care and Customer Support
Electrostatic Discharge (ESD), Handling Notice…… 8-1
Product Care …… 8-1
The Readme Files and the Install CD-ROM …… 8-2
Customer Support …… 8-2

Chapter 9 – Specifications, IOtech 6224

1

For information regarding digital circuitry, refer to the Digital I/O section of the Specifications chapter.

6224 User’s Manualvii

IOtech 6224

Quick Start

High-speed, Strain Gauge Input Module
Before you get started verify that you have the following items
and meet or exceed the minimum requirements listed.
IOtech 6224
TR-60U Power Supply and Cable
Ethernet Crossover Cable*
IOtech Encore Installation CD
PC: Minimum Requirements:
CPU: Intel® Pentium® 4, 3.0 GHz or equivalent
RAM: 1 GB
Monitor: 1024 by 768 screen resolution
PC: Recommended:
CPU: Intel® CoreTM 2 Duo Family
RAM: 2+ GB
Monitor: 1024 by 768 screen resolution*An Ethernet crossover cable (provided) should only be used when connecting a 6000
Series device directly to the PC. A standard Ethernet cable should be used whenconnecting via a hub/switch. A Gigabit switch is recommended when connecting
multiple devices.

Step 1 - Connect the 6224 to the Ethernet and to the Power Supply
IMPORTANT: You must use a crossover cable ifconnecting a 6000 Series device directly to the PC.

In this scenario the 6224 is connected directly to an Ethernet port on a host computer.
1.

Connect the Ethernet cable to the Ethernet jack on the 6224. If connecting directly to the PC, as indicatedabove (i.e., instead of connecting to a network hub) you must use an Ethernet crossover cable.

2.

Connect the other end of the Ethernet cable to the Ethernet jack on the host computer or network hub.

3.

Connect the power supply cable from the TR-60U Power Supply to the External Power connector on the 6224.

4.

Connect the TR-60U to a standard AC outlet; and set the rear panel Power Switch to “I” to turn the unit on.
The 6224 Power LED will light up.

5.

If Encore software is already installed, reboot the PC and skip to Step 3; otherwise proceed to Step 2, Install
Encore Software.

Step 2 – Install Encore Software
1.

Close all running applications on the host PC.

2.

Insert the Encore CD into your CD-ROM drive and wait for the CD to auto-run.
An Opening Screen will appear. If AutoRun is disabled: (a) right-click Windows Start menu,(b) select Explore, (c) navigate to the CD ROM drive, and (d) double-click Autorun.exe

3.

Follow the screen prompts.

Rev 1.2

325130C-01

Step 3 – Use Encore to Configure the 6224
1. Run the Encore software application (Start>>All Programs>>IOtech>>Encore).
2. In Encore’s main toolbar, click the Hardware Configuration button. The Hardware Configuration window will open.
3. Select your device from the Detected Devices panel.
4. Click the top arrow button to set your device as a Project Device.
5. In Encore’s Hardware Configuration window, look in the Device Information panel to verify that your 6224 has an
IP Address. Note the following points regarding common connection scenarios. Both points assume that you have notchanged the 6224 from its default setting of Obtain an IP Address Automatically.
oDirect Connect – When a 6000 Series device is connected directly (via crossover cable or hub/switch) to a host PC
with no fixed IP Address, you will need to reboot the PC and the device to allow a default IP Address to be set. Thissetting of an IP Address is automatic. No other action is needed. (See note.)oNetwork Connect (Common scenario using existing network with DHCP server) – A 6000 Series device willautomatically request and receive an IP Address. No user action is needed. (See note.)

Note - Encore will provide a message to indicate an IP Address problem if one exists. If you receive such a message,
consult your network administrator and also check Encore’s Comprehensive Help file (via the Help pull-down menu on
Encore’s main window). Look at the topic: Connection Scenarios and IP Addresses (under Configuring Hardware>>Hardware Setup).
6. Click the Next button (at the bottom of the Hardware Configuration dialog) to access the Acquisition Setup tab.
7. Configure and enable channels as desired for your application. For detailed information refer to Encore’s Comprehensive
Help file (via the Help pull-down menu on Encore’s main window).
Make sure that each computer and each device on the dedicated network has a unique IP address.

Step 4 - Connect Signal Lines
CAUTION
Turn off power to the system devices and externally connected equipment before connecting
cables. Electric shock or damage to equipment can result even under low-voltage conditions.
Take ESD precautions (packaging, proper handling, grounded wrist strap, etc.)
The 6224 uses 10-pin RJ50 jacks for strain connection. Do not connect RJ45 plugs to these
RJ50 jacks. Using RJ45 plugs can cause permanent damage to RJ50 pins 1 and 10, thus
disabling shunt calibration.

The 6224 has 12 analog channel inputs (CH1 through CH12) via front panel RJ50 jacks and 8 digital I/O lines via
rear panel DB9 connector.
Prior to making signal connections review the Specifications chapter of your 6224 user’s manual to ensure that the input
signals do not exceed the specified limits. After you have installed Encore you can access the pdf manual by navigatingfrom the host PC’s Windows Start Menu as follows (location shown is the default):
Start>>All Programs>>IOtech>>6000 Series>>6224 User Manual
Reference Note: The Encore software application includes Context-sensitive Help, Comprehensive Help,a Quick Start Video Tutorial, and a Getting Started pdf.

Measurement Computing 10 Commerce Way, Norton, MA 02766
Phone: (508) 946-5100, Fax: (508) 946-9500 E-mail: info@mccdaq.com
Internet: www.mccdaq.com*325130C-01*
325130C-01

Printed in Hungary

General Information

1

Introduction …… 1-1
Front Panel …… 1-1
Rear Panel …… 1-2
Analog Circuitry1 …… 1-3
Synchronization …… 1-4
Stacking Plate and Handle Options ….. 1-5
Quarter-Bridge Completion Accessories(CN-269 & CN-270) …… 1-6
RJ50 12-Pin Screw Terminal Accessory (CN-268)….. 1-7
RJ50 Cables (CN-272-01) …… 1-7

Introduction
The 6224 is a twelve-channel, 24-bit resolution strain measurement device that connects to a host PC‟s Ethernet
port. The device is used to measure half-bridge and full-bridge sensors and can be synchronized with other 6000
series devices.
The hardware provides the signal conditioning and acquisition engine, while the software defines the specificconfiguration and analysis aspects of the system.
Note that the software includes both a context sensitive Help and a comprehensive Help file.

6224 Basic profile:
Model

6224

Interface

10/100BaseT
Ethernet

Analog
Input
Channels

12 Strain
RJ50 Jacks

SYNC
Ports

2

AnalogOutputChannels0

DigitalI/O

1 port/8 lines

TEDS
SupportSupported Bridges(Note 3)(Note 2)

Quarter-Bridge I
Quarter-Bridge II
Half-Bridge I
Half-Bridge II
Full-Bridge I
Full-Bridge II
Full-Bridge III

Front Panel
The front panel houses twelve channel signal input jacks andthree excitation voltage source connectors, one for each set offour channels.
The input channel jacks are type RJ50 (10p10c) and are labeled
CH1 through CH12. The 10-connector pinout for the RJ50
connectors is included in chapter 3.
Optional external voltage source connectors are available. For
additional information see the External Excitation Voltage
Sources section of chapter 3.

Do not connect an RJ45 plug to an RJ50 jack. Using RJ45 plugs can cause
permanent damage to RJ50 pins 1 and 10, thus disabling shunt calibration.

1

For information regarding digital circuitry, refer to the Digital I/O section of the Specifications chapter.
TEDS is supported by the 6224, but only if TEDS is supported by the software application.
3
For information regarding the various bridge types, refer to chapter 7.
26224 User’s Manual

General Information

1-1

Rear Panel
The following items reside on the rear panel; fromleft to right: 1st cable tie mount, Ethernet jack,DB9-Digital I/O port, 2nd cable tie mount, #6-32
chassis ground screw, four status LEDs, two
synchronization jacks, a power input jack, and apower switch. A description of each follows.
CABLE TIE MOUNTS: Two cable tie mounts are present to provide for strain relief of cables.
ETHERNET: The 10/100BaseT Ethernet port can connect to the Ethernet port of the host PC, or to an Ethernet
network. The Ethernet connector has two built in LEDs that indicate traffic flow on the network.
Refer to Accessories in the Specifications chapter for Ethernet cables available from Measurement Computing.
When connecting the unit directly to a PC (i.e., when not connecting to a network hub) you
will need to use an Ethernet crossover cable.
Ethernet cable length must be <3 m (9.8 ft) in order for the system to be CE Compliant.
DIGITAL I/O: Eight digital I/O lines are accessible via a 9-pin, female DSub connector. For details, refer to theDigital I/O section of the Specifications chapter.
GROUND (#6-32): Provides a connection point for Chassis Ground via use of a #6-32 machine screw.
STATUS LEDS:
Power„ON‟ when the device is connected to a sufficient power source and the power switch isin the „l‟ (On) position.

Boot

During initial power up, the Boot light will be On to signal the first stage boot process.
When this has completed successfully, the Boot light will remain Off. If an internal
error is detected during this first stage boot process, this light will be blinking. Contact
Measurement Computing to arrange repair service.

ActiveDuring initial power up, the Active light will be Off during the boot process, including
network configuration. The process will complete with a fast Blink, and then steady On
condition. With both the Power and Active lights On, the device is in the Ready state.
After this, the Active light will signal communications with a host computer at a slower
blink rate.

Data

The Data light will be On any time the device is acquiring channel data that is available
to the host PC control software.

Note:

After turning on the device, it can take up to 2 minutes to reach ready state, depending on
network settings. This can be monitored by the LED sequence:
Power Light On Boot Light On then Off Active Light Off, then Blinking, then On.

SYNC: Two “synchronization ports” provide a means of synchronizing units regard to pre-trigger data and posttrigger scanning. Note that each 6000 Series device is capable of being designated as a Master Unit, Slave Unit, or
Terminating Slave Unit. The assignment is made in Encore. Refer to the following section, Synchronization, fordetails. Refer to Accessories in the Specifications chapter for SYNC cables available from Measurement Computing.
Up to nine units can be synchronized. The total combined length of the SYNC cables is not
to exceed 2.438m (8 ft).
POWER INPUT: +19 VDC to +30 VDC, through a connector on the chassis. Power is typically supplied to thedevice via a TR-60U, 24VDC universal power supply.
POWER SWITCH: A switch with a “0” label for Power Off, and a “ ” for Power On.

1-2

General Information

6224 User’s ManualAnalog Circuitry
The 6224 is isolated from earth ground. However, the individual channels, within their four-channel group, are notisolated from each other. The EX+, EX–, and T– signals are common among the four channels in their group as
indicated by note 1 on page 3-1 and in the block diagram (page 2-1).
You can connect the 6224 to a device that is biased at any voltage within the unit‟s rejection range of earth ground.
The Specifications chapter includes additional information.

Filtering

Passband -Stopband-

Alias-Free Bandwidth -6224 User’s Manual

General Information

1-3

Synchronization
To synchronize from two to nine 6000 Series devices, linkthem together with synchronization cables as indicated in theexample figure. Note that the first device (relative to thecircuit) will be the Master. The last device connected will be
the Terminating Slave. Thus the synchronized devices must
proceed sequentially from the Master to Terminating Slave, as
indicated in the figures. Note that channel-to-channel phaserelationships in a multi-device configuration are notnecessarily fixed. See notes below.
After physically linking the devices via the sync cables, they must still be configured as a “Synchronized Device
Group” in the Encore software. To do this, use the Synchronize Devices page of Encore‟s Hardware Setup tab.

Each Synchronized Group consists of one Master Device
and at least one Slave Device. By default, the first 6000
Series device added to the Synchronization Group will be
the Master and the last device added to the group will be theTerminating Slave.

TO ACQTO ACQ

Encore‟s Synchronized Device Group Configuration
panel includes a pull-down list which allows you to selectthe device that you want to have function as Master and theone you want to function as Terminating Slave, should thisbe different than the default settings.
Encore will display a Synchronization Error icon if it
detects that a necessary synch cable is missing or notproperly connected.

TO ACQTO ACQ

Up to nine units can be synchronized. The total combined length of the SYNC cables is not
to exceed 2.438m (8 ft).
Unplug sync cables from devices that are not intended to be part of a “Synchronized Device
Group.” Leaving a sync cable connected to an independent device can lead to errors.
Leaving an un-terminated sync cable on a synchronized device can lead to errors.
The sampled data phase relationship among channels between multiple devices isdependent on the Channel Sync Skew specification for each device. When utilizing
different models in a multi-device system, any differences in the ADC filter delay(“Input Delay”) should be added.

Reference Note: For detailed information refer to the Encore software‟s Comprehensive Help.
You can access the Help from within Encore. After doing so, navigate to the synchronization
related topic as follows from the Help‟s Contents panel:
Configuring Hardware>>Hardware Setup>>Synchronizing Devices.

1-4

General Information

6224 User’s ManualStacking Plate and Handle Options
6000 Series device chassis are equipped with feet (on the lower panel) and dimples (on the upper panel) to allow for
quick stacking of units.
A rigid securing of two or more units together is possible with the use of optional stacking plate kits (190658A-01).
Each kit contains two stacking plates and eight screws (8-32 x .500 in., Phillips Flat, 82 Degree). If needed,
additional kits can be used to stack and secure more devices.
An optional handle kit (HA-210-5-BK) is available to provide a convenient way of carrying a single 6000 Series
device or a secured stacked set of devices. Each handle kit includes one black molded plastic handle and two
mounting screws (1/4-20 x 7/8 in., Phillips Pan Head).

6000 Series Devices

HandleStacking Plate (1 of 2)(from kit HA-210-5-BK)(from kit 190658A-01)

To attach two or more 6000 Series devices to each other using stacking plates:
1.

Align the bottom two holes of a plate (hole tapers facing out) with the top two center holes of the lower
6000 Series device.

2.

Secure the plate with two 8-32 x .500 in. Phillips screws (provided). Tighten the screws snug, but do not
over-tighten.

3.

Repeat steps 1 and 2 for the second plate, on the other side of the 6000 Series device.

4.

Attach the second 6000 Series device using the remaining four screws.

5.

Repeat these steps with other stacking plate kits and devices as applicable.

To mount a handle:
Attach the handle to a 6000 Series device using the two outer holes on the device and the two 1/4-20 x 7/8 in.,
Phillips Pan Head screws (provided). Tighten the screws snug, but do not over-tighten.

6224 User’s Manual

General Information

1-5

Quarter-Bridge Completion Accessories (CN-269 & CN-270)

The CN-269 and CN-270 are Quarter-Bridge Completion kits, with each kit
consisting of four components. A single component (each member of its
respective four-pack) can be used to add a Quarter-Bridge completion resistor
to a channel on the 6224.
The CN-269 components each house a 120 ohm resistor.
The CN-270 components each house a 350 ohm resistor.
You will need an RJ50 M/M cable to use any of the Quarter-Bridgecomponents with an 6224. Measurement Computing offers a pack of four
RJ50 M/M, 1 meter (3.28ft) cables (see upcoming RJ50 Cables section).

As indicated in the right-hand figure below, CN-269 and CN-270 components each have two holes that can be used to loop aplastic tie-strap through, as a means of adding strain relief for the wire connections to the screw terminals.
If you want to mount a component, use a M4x20 screw (or similar hardware) that can fit into the 8.89 mm (0.350 in.) outer
clearance/4.57 mm (0.180 in.) inner clearance hole. See figure.

Specifications(typical for the range -40 to 70 °C)
Weight ................................................ 32 g (1.1oz) per single component
Operating temperature .................... –40 to 70 °C
Resistor values:
CN-269 .......................................... 120 Ω
CN-270 .......................................... 350 Ω
Resistance drift:
CN-269 .......................................... 0.012 Ω/°C
CN-270........................................... 0.035 Ω/°C
Resistor tolerance (at 25 °C) ............... 0.1% max
Resistor power .....................................0.25 W max

Do not connect an RJ45 plug to an RJ50 jack. Using RJ45 plugs can cause
permanent damage to RJ50 pins 1 and 10, thus disabling shunt calibration.

1-6

General Information

6224 User’s ManualRJ50 12-Pin Screw Terminal Accessory (CN-268)
This accessory is used for Half-Bridge or Full-Bridge completion. Because bridge
completion is built into the 6224, you will only need connectivity accessories.
The CN-268 is a set of four female RJ50 to 12-pin screw terminal adapters. You
will need an RJ50 M/M cable to use a CN-268 component with an 6224.
Measurement Computing offers a pack of four RJ50 M/M, 1 meter (3.28ft) cables(see following RJ50 Cables section).

As indicated in the right-hand figure below, CN-269 and CN-270 components each have two holes that can be used toloop a plastic tie-strap through, as a means of adding strain relief for the wire connections to the screw terminals.
If you want to mount a CN-268 component, use an M4x20 screw (or similar hardware) that can fit into the 8.89 mm(0.350 in.) outer clearance/4.57 mm (0.180 in.) inner clearance hole. See figure.

Specifications:
Weight ...................................................... 43 g (1.5oz)
Operating temperature ............................ –40 to 70 °C

Do not connect an RJ45 plug to an RJ50 jack. Using RJ45 plugs can cause
permanent damage to RJ50 pins 1 and 10, thus disabling shunt calibration.

RJ50 Cables (CA-272-01)
When connecting a cable from an 6224 RJ50 jack to an associated Measurement Computing accessory (CN-268, CN-269,
or CN-270), be sure to only use RJ50 M/M cables. Use of another cable type can damage RJ50 pins.
Pin 10

Pin 1

P1

P2

1 m (3.28ft)

The CA-272-01 includes four RJ50 to RJ50 M/M, 1 meter (3.28ft) cables.

6224 User’s Manual

General Information

1-7

Block Diagram

6224 User’s Manual2

Block Diagram

2-1

Pinouts and Connections

3

Strain Channels, RJ50 Pinout …… 3-1
Bridge Connections …… 3-2
Connecting TEDS Channels …… 3-2
Remote Sensing …… 3-2
Shunt Calibration …… 3-3
Internal Excitation Voltage Sources …… 3-3
External Excitation Voltage Sources …… 3-4
Mounting and Securing Connections …… 3-5
Where to Find Additional Information …… 3-5

Strain Channels, RJ50 Pinout
The twelve RJ50 (10p10c) jacks each have 10 pins with signal designations as indicated in the following table.

RJ50 (10p10c) Modular Plug and Jack Pinout
Pin #

RJ50
Front View

Signal Name

Signal Description

1

SC

Shunt Calibration

2

AI+

Positive Input Signal

3

AI-

Negative Input Signal

4RS+

Positive Remote Sense Signal

5RS-

Negative Remote Sense Signal

6EX+

Positive Excitation Signal (Note 1)

7

EX-

Negative Excitation Signal (Note 1)

8

T+

TEDS DATA (Note 2)

9

T-

TEDS Return (Notes 1 & 2)

10

SC

Shunt Calibration

Note 1: The three channel blocks each share three signals [within their block], as follows:
For CH1, CH2, CH3, and CH4, the signal values of pins 6, 7, and 9 are shared.
For CH5, CH6, CH7, and CH8, the signal values of pins 6, 7, and 9 are shared.
For CH9, CH10, CH11, and CH12, the signal values of pins 6, 7, and 9 are shared.

Note 2: The 6224 only supports TEDS if the acquisition software being used supports TEDS.

Do not connect an RJ45 plug to an RJ50 jack. Using RJ45 plugs cancause permanent damage to RJ50 pins 1 and 10, thus disabling shuntcalibration.

6224 User’s Manual

Pinouts and Connectors

3-1Bridge Connections
The following figures indicate how to connect bridges to an the 6224. The manner in which thesesignals correspond to the RJ50 connector is shown on the preceding page.

Quarter-Bridge

Half-Bridge

Full-Bridge

Notes:(1) When connecting a bridge to the 6224, the following connections are optional: RS+, RS-, SC, T+, T-.
(2) For the half-bridge and quarter-bridge, there is no AI- signal to connect.
(3) To create a quarter-bridge, place an external resistor across the two SC lines of a half-bridge. Measurement
Computing offers two external Quarter-Bridge accessories, CN-269 (120 ohm) and CN-270 (350 ohm). See
chapter 1 for additional information regarding these accessories.
(4) Chapter 7 contains additional bridge information, including diagrams of the various bridge types that can beselected in Encore and whether they apply to axial and/or bending strain.

Connecting TEDS Channels
Before connecting TEDS, be sure that your software supports TEDS. If it does, when connecting
TEDS ensure that neither TEDS data (T+) nor TEDS return (T–) is tied in common to any AI signals
on the 6224. See preceding schematics.

Remote Sensing
Wire resistance can create errors in bridge circuits. The
6224 can correct for these errors using two mechanisms: (1)Remote Sensing and (2) Shunt Calibration.
Remote sensing continuously and automatically corrects forerrors in excitation leads, and generally is most appropriate
for full- and half-bridge sensors.
Long wire and smaller gauge wire have greater resistance,
which can result in gain error. The voltage drop caused bywire resistance in the wires, which connect the excitation
voltage to the bridge, is a source of gain error. The 6224
includes remote sensing to compensate for this error.
Remote sense (RS) wires are connected to the point where
the excitation voltage wires connect to the bridge circuit, asseen in the figure.

3-2

Pinouts and Connections

6224 Users ManualIn the preceding figure, the actual bridge excitation voltage is smaller than the voltage at the EX+ and EX– leads.
If remote sensing of the actual bridge voltage is not used, the resulting gain error is as follows:
Error in Full-bridge sensors = 2RLead/RBridge
Error in Half-bridge sensors = RLead/RBridge
If the remote sense (RS) signals are connected directly to the bridge resistors, then the 6224 senses
the actual bridge voltage sense and eliminates the gain errors caused by the resistance of the EX+ and EX– leads.

Shunt Calibration
The second mechanism [aside from remote sensing] used to correct errors from wire resistance is Shunt
Calibration. This involves simulating the input of strain by changing the resistance of an arm in the bridge.
This is accomplished by shunting, or connecting, a large resistor of known value across one arm of thebridge, creating a known strain-induced change in resistance. The output of the bridge can then bemeasured and compared to the expected voltage value. The results are used to correct gain errors in theentire measurement path, or to simply verify general operation to gain confidence in the setup.
Shunt calibration can be used to correct for errors from the resistance of both the excitation wiring andwiring in the individual resistors of the bridge. Shunt calibration is most critical when measurements are
made on Quarter-Bridge sensors because there is no means to remotely sense around any IR drops in theconnection wiring.
The 6224 includes a precision 100kΩ resistor and a software-controlled switch for each channel. You canleave the shunt calibration terminals connected to the sensor, and then apply or remove the shuntcalibration resistance in software.
While remote sensing corrects for resistances from the EX terminals on the 6224 to the sensor, shuntcalibration corrects for these errors and for errors caused by wire resistance within an arm of the bridge.
A stable signal, which is typically the unloaded state of the sensor, is used first with the shunt calibration
switch OFF, and then again with the switch ON. The difference in these two measurements provides anindication of the gain errors from wiring resistances and corrections for offset errors.

Internal Excitation Voltage Sources
The 6224 houses three internal voltage sources; each serves 4 channels. The first source applies excitation voltage to
CH1 through CH4, the second to CH5 through CH8, and the third to CH9 through CH12.
The sensor industry does not recognize a single standard excitation voltage level. However, excitation voltage levels
residing in the range of 2.5V to 10 V are common. Encore’s selections for internal excitation are 2.5V, 3.3V, 5V,
and 10V, and each of the three internal voltage sources can provide up to 150 mW of excitation power. The 6224
automatically reduces internal excitation voltages, as needed, to stay below 150mW.
Since channels are associated by groups of four, the excitation setting applied to one channel in a group will also be
applied to the other channels in that group. As an example: if channel 1 is set to have and internal excitation of 3.3V,
then channels, 2, 3, and 4 will also have 3.3V excitation.
The 6224 measures the ratio of bridge output to bridge excitation, and as such does not require accurate excitation
voltage. For this reason the excitation voltage is not precisely regulated and may vary as much as
10% from the requested voltage, while still achieving accurate measurements.

6224 User’s Manual

Pinouts and Connectors

3-3

2

The power consumed by a single bridge is Vex /R
Where Vex is the excitation voltage and R is the total resistance of the bridge.

For a full bridge, R is equal to the resistance of each element. For a half or quarter bridge, R is equal to two times
the resistance of each element.
For each of the three voltage sources, the 150 mW limit allows you to power full and half bridges as follows:
Four 350 Ω half bridges at 5.0 V, or
Four 350 Ω full bridges at 3.3 V, or
Four 120 Ω half bridges at 2.5 V

If you need an excitation voltage greater than 150 mW, use the four-position external excitation voltage connectors(see following section). Each of the three external connectors can be used to provide external excitation to fourchannels of the 6224; thus all twelve channels can receive external excitation.

External Excitation Voltage Sources
The 6224 front panel includes three 4-pin connectors which provide a means of connecting external excitation
voltage sources.
Each external voltage source serves 4 channels, as follows; the first source applies excitation voltage to CH1
through CH4, the second to CH5 through CH8, and the third to CH9 through CH12. Since channels are associatedby groups of four, what you set for one channel (in a group) will also be applied to other channels of that group.
Example: if you set channel 1 for 10V external excitation, channels, 2, 3, and 4 will also have 10V external
excitation.
The sensor industry does not recognize a single standard excitation voltage level. However, excitation voltage levels
residing in the range of 2.5V to 10 V are common.
Unless you supply external excitation voltage, we recommend that you set the excitation voltage
to a value that keeps the total power below 150 mW.

The pinout and manufacturing information for the external excitation voltage source connector follows.
Contact Measurement Computing for more information.
Top of Chassis-+

Silkscreen polarity orientation
markers visible on 6224 chassis

Manufacturer Information
Molex Mating Connector
4-position housing (43025-0400)
Crimp terminal (43030-0007)

Also see the preceding section, Internal Excitation Voltage Sources.

3-4

Pinouts and Connections

6224 Users ManualMounting and Securing Connections
For a strain measurement system "mechanical noise" should also be a consideration. Movement in theconnection system could translate to errors in the measurement.
Care should be taken to ensure the 6224, RJ50 cables, and dongles are physically secured to prevent
relative movement.

Where to Find Additional Information
For detailed information, refer to the applicable software documentation, including software help files. Notethat the Encore software includes both a Context-sensitive Help and access to a Comprehensive Help
document.
In addition to software documentation, information regarding aliasing, anti-aliasing, sensors, accelerometers,sound and vibration, and transducers may prove useful to your application. The internet is a great source forthis material.
In regard to TEDS (Transducer Electronic Data Sheet), you may find the IEEE 1451 standard to beof interest.

6224 User’s Manual

Pinouts and Connectors

3-5Counter Input

4

The 6224 has four 32-bit counter inputs that accept frequency inputs up to 20 MHz. The counter inputs areindividually configurable for counter or encoder measurements.
You activate each input, enable logging, and configure counter options with Encore.

Counter Settings
Counter Source

The source to use for the counter. Choices are:
 Internal Clock Timer 1
 Timer 2
 Digital Line 0 – Digital Line 7
One source can be used in multiple counters.

Counter Gate

The gate to use for the counter. Choices are:
 Unused (do not gate the counter) Internal Clock Timer 1
 Timer 2
 Digital Line 0 – Digital Line 7
One gate can be used in multiple counters.

Count Settings

The functions or values for the counter source. Choices are:
 Clear on Read – The counter counts up and is cleared after each read. By default,
the counter counts up and only clears the counter at the start of a new scancommand. The value of the counter before it is cleared is latched and returned.
 Totalize – The counter counts up and is cleared at the start of a new scan.
 Start Count value – The value used to start counting. The default is zero.
 Stop Count value – The value used to stop counting. The default is 65535.
 Rollover – The count rolls over upon reaching the start or stop value.

Counter Source
Direction

The direction in which to count. Choices are:Counter Source
Detection

The count to detect in relation to the signal edge. Choices are:6224 User’s Manual Increment – Count up (default).
 Decrement – Count down.

 Rising Edge – Detects the count when the signal goes from low to high.
 Falling Edge – Detects the count when the signal goes from high to low.

Counter Input

4-1

Encoder Settings
Resolution

The encoder resolution, which is the number of full quadrature cycles per full shaft
revolution (360 mechanical degrees). Choices are: X1 – One count per cycle (default)
 X2 – Two counts per cycle
 X4 – Four counts per cycle

Encoder Sources

There are three encoder sources – A, B, or Z. For each source select one of thedigital pins (Digital Line 0 – Digital Line 7)

When set for Encoder measurements, the device can count "negative" values;that is, when counting down, the encoder will continue counting down below 0.
This capability is useful when calculating the position of an encoder.

4-2

Counter Input

6224 User’s ManualCE Compliance & Noise Considerations

5

Overview …… 5-1
Safety Conditions …… 5-1
Emissions/Immunity Conditions …… 5-2
CE Rules of Thumb …… 5-2
Noise Considerations …… 5-3

Overview
CE compliant products bear the “CE” mark and include a Declaration of Conformity stating theparticular specifications and conditions that apply. The test records and supporting documentationthat validate the compliance are kept on file at the factory.
The standards are published in the Official Journal of European Union under direction of CENELEC(European Committee for Electrotechnical Standardization). The specific standards relevant to dataacquisition equipment are listed on the product’s Declaration of Conformity.
This product meets the essential requirements of applicable European directives, as amended for
CE markings in accordance with the product family standard for:electrical equipment for measurement, control, and laboratory use
immunity requirements for equipment used in controlled EM environments
Refer to this product’s Declaration of Conformity (DoC) for any additional regulatory complianceinformation. You can view our Declarations of Conformity at www.mccdaq.com/legal.aspx (CEInformation page).

Safety Conditions
Users must comply with all relevant safety conditions as stated in the user’s manual and in the pertinent
Declarations of Conformity. Both the documentation and the associated hardware make use of thefollowing Warning and Caution symbols. If you see any of these symbols on a product or in a document,
carefully read the related information and be alert to the possibility of personal injury and/or equipment
damage.
This WARNING symbol is used in documentation and/or on hardware to warn ofpossible injury or death from electrical shock under noted conditions.
This WARNING/CAUTION symbol is used to warn of possible personal injury or
equipment damage under noted conditions.
This CAUTION symbol warns of possible equipment damage due to electrostatic
discharge. The discharge of static electricity can damage some electronic
components. Semiconductor devices are especially susceptible to ESD damage. You
should always handle components carefully, and you should never touch connector
pins or circuit components unless you are following ESD guidelines in an appropriate
ESD-controlled area. Such guidelines include the use of properly grounded mats andwrist straps, ESD bags and cartons, and related procedures.
Unless otherwise stated our data acquisition products contain no user-serviceableparts. Only qualified personnel are to provide service to the devices.

User’s Manual

CE-Compliance & Noise Considerations

5-1

The specific safety conditions for CE compliance vary by product; but general safety conditions include thefollowing bulleted items:
The operator must observe all safety cautions and operating conditions specified in the
documentation for all hardware used.
The host computer and all connected equipment must be CE compliant.
All power must be off to the device and externally connected equipment before internal access to thedevice is permitted.
Ensure that isolation voltage ratings do not exceed documented voltage limits for power and signal
inputs. All wire insulation and terminal blocks in the system must be rated for the isolation voltage
in use. Voltages above 30 Vrms or ±60 VDC must not be applied if any condensation has formed on
the device.
Current and power use must not exceed specifications. Do not defeat fuses or other over-current
protection.

Emissions/Immunity Conditions
The specific immunity conditions for CE compliance vary by product. General immunity conditions include thefollowing:
Cables must be shielded, braid-type with metal-shelled connectors. Input terminal connections are to bemade with shielded wire. The shield should be connected to the chassis ground with the hardware provided.
The host computer must be properly grounded.
In low-level analog applications some inaccuracy is to be expected when I/O leads are exposed to RF fields
or transients, as noted on the Declaration of Conformity, if applicable to the device.

CE Rules of Thumb
The device is CE Compliant at the time it leaves the factory and should remain in compliance as long as theconditions stated on the Declaration of Conformity continue to be met.
A few general rules of thumb:
Use short cables.
When assembling or disassembling components, take ESD precautions,
including the use of grounded wrist straps.
Ensure that the host computer is CE Compliant.
Review the most recent Declaration of Conformity.
Ensure all system components are properly grounded.

5-2

CE-Compliance & Noise Considerations

User’s Manual

Noise Considerations
Controlling electrical noise is imperative because it can present problems even with the best measurement
equipment. Most laboratory and industrial environments suffer from multiple sources of electrical noise.
For example, AC power lines, heavy equipment (particularly if turned on and off frequently), local radio
stations, and electronic equipment can create noise in a multitude of frequency ranges.
Local radio stations are a source of high frequency noise, while computers and other electronic equipment
can create noise in all frequency ranges. Creating a completely noise-free environment for test andmeasurement is seldom practical. Fortunately, simple techniques such as using shielded/twisted pair wires,
filtering, and differential voltage measurement are available for controlling the noise in our measurements.
Some techniques prevent noise from entering the system; other techniques remove noise from the signal.
While many techniques for controlling noise in signals provide a means of removing the noise that isalready present, the preferred solution is to prevent the occurrence of noise in the signal in the first place.
The following practices, some of which are required for CE compliance, should be employed to minimize
noise.
Make a solid earth ground connection. Using insulated, low resistance wire, connect thechassis to solid earth ground. This practice accomplishes the following:(a) keeps radiated emissions low by keeping the chassis electrically quiet,(b) keeps potential common-mode voltages low,(c) improves user safety, and(d) provides a safe path for Electrostatic Discharge energy back to earth ground.
Minimize ambient EMI. The lower the ambient EMI, the better. Sources of electromagnetic
interference include solenoids, motors, computer equipment, high power distribution wiring,etc.
Distance cables. Route signal cables away from Ethernet lines, Mains and other high voltage
cables and equipment to minimize signal interference from radiated noise.

User’s Manual

CE-Compliance & Noise Considerations

5-3

Software, IOtech Encore

6

Encore is the premier Out-of-the-Box data acquisition software package and is included with each 6000 Series
module. Encore combines ease of use with advanced functionality, including customizable data layouts, powerful
analysis, and reporting capabilities. Instead of having a program to log data, another to analyze, and a third todevelop report data, Encore includes the functionality of all three into one package, thus shortening the learning
curve, and saving time and cost.

Features
Easily configure 6000 Series hardware
Data streaming & playback
Comprehensive report generation
Drag and drop directly into Excel®
Built-in analysis including FFT

We recommend that you watch the Encore Quick Start Video Tutorial prior to using Encore. After watching thevideo you should be able to set up and run an application. You can use Encore’s Comprehensive Help to view
details about various aspects of the application. Comprehensive Help includes a search function, allowing you toquickly look up the subject of interest.
Once you have installed Encore you can navigate from the host PC’s Windows’ Desktop to access theComprehensive Help and several other items as indicated in the table below.
Use this Path from the Desktop:

To Access:File Format

Start>>All Programs>>IOtech>>Encore>>Comprehensive Help
EncoreGetting Started with Encore
Quick Start Video Tutorial
ReadmeAll 6000 Series user’s manuals.chm
software application
.pdf
.wmv
.html
.pdf

Start>>All Programs>>IOtech>>6000 Series>>

Should you have questions or need support you can contact Measurement Computing through one of the means
provided on the front cover of this manual.

6000 Series

Software6-1

Strain and Pressure/Force

7

Introduction …… 7-1
Strain Gage Types Explained …… 7-2
Quarter-Bridge I …… 7-3
Quarter-Bridge II …… 7-4
Half-Bridge I …… 7-5
Half-Bridge II …… 7-6
Full-Bridge I …… 7-7
Full-Bridge II …… 7-8
Full-Bridge III …… 7-9

Pressure/Force …… 7-10

Introduction
The 6224’s channels can be set to measure strain or pressure/force. This chapter provides schematics of bridge
circuits applicable to the 6224.

6224 User’s Manual

Strain and Pressure/Force

7-1

Strain Gauge Types ExplainedThis section uses figures to represent axial and/or bending strain to indicate the type(s) of strain that applies to agiven bridge type. It also provides simple schematics. For each type of bridge, supplemental information shows
two equations: the first is the equation used to convert voltage to strain; the second is an equation you can use tosimulate the effect on strain by applying a shunt resistor across R3. In the upcoming figures and equations,acronyms, formulas, and variables are defined as:is the measured strain (+ is tensile strain and -is compressive strain).

S is the simulated strain.

GF is the Gauge Factor, which should be specified by the gauge manufacturer.
Rg is the nominal gauge resistance, which should be specified by the gauge manufacturer.
RL is the lead resistance. If lead lengths are long, RL can significantly impact measurement accuracy.
Rs is the shunt calibration resistor value.
U is the ratio of expected signal voltage to excitation voltage with the shunt calibration circuit engaged.
Parameter U appears in the equations for simulated strain and is defined by the following equation:
Rg
U

4Rs

2Rg

is the Poisson’s ratio, defined as the negative ratio of transverse strain to axial strain (longitudinal) strain.
VCH is the measured signal’s voltage.
VEX is the excitation voltage.
Vr is the voltage ratio that is used in the voltage to strain conversion equations and is defined by thefollowing equation:

Vr

VCH (strained)

VCH (unstrained)VEX

7-2 Strain and Pressure/Force

6224 User’s ManualQuarter-Bridge I
Quarter-bridge type I measures either axial or bending strain.

Quarter-bridge type I has the following characteristics:
A single active strain-gauge element is mounted in the principle direction of axial or bending strain.
A passive quarter-bridge completion resistor (dummy resistor) is required in addition to half-bridge completion.
Temperature variation in specimen decreases the accuracy of the measurements.
Sensitivity at 1000is ~ 0.5 mVout/ VEX input.
The following symbols apply to the above circuit diagram and to the equations provided in the supplemental information.
R1 and R2 are the half-bridge completion resistors.
R3 is the quarter-bridge completion resistor (dummy resistor).
R4 is the active strain-gauge element measuring tensile strain (+ ).

Supplemental Information
Data from strain channels (Quarter-Bridge I) is converted to strain units using the following equation:
4 Vr

6

10

GF 1

6224 User’s Manual

2 Vr

1

RL
Rg

Strain and Pressure/Force

7-3

Quarter-Bridge II
Quarter-bridge type II measures either axial or bending strain.

Quarter-bridge type II has the following characteristics:
One active strain-gauge element and one passive, temperature-sensing quarter-bridge element (dummy gauge). The
active element is mounted in the direction of axial or bending strain. The dummy gauge is mounted in close thermal
contact with the strain specimen but not bonded to the specimen, and is usually mounted transverse (perpendicular)
to the principle axis of strain.
This configuration is often confused with the half-bridge type I configuration, with the difference being that in thehalf-bridge type I configuration the R3 element is active and bonded to the strain specimen to measure the effect ofPoisson’s ratio.
Completion resistors provide half bridge completion.
Compensates for temperature.
Sensitivity at 1000is ~ 0.5 mVout/ VEX input.
The following symbols apply to the above circuit diagram and to the equations provided in the supplemental information.
R1 and R2 are half-bridge completion resistors.
R3 is the quarter-bridge temperature-sensing element (dummy gauge).
R4 is the active strain-gauge element measuring tensile strain (+ ).
Supplemental Information
Data from strain channels (Quarter-Bridge II) is converted to strain units using the following equation:

6

10

4 Vr
GF 1

2 Vr

7-4 Strain and Pressure/Force

1

RL
Rg

6224 User’s ManualHalf-Bridge I
Half-bridge type I measures either axial or bending strain.

Half-bridge type I has the following characteristics:
Two active strain-gauge elements. One is mounted in the direction of axial strain, the other acts as a Poisson gauge
and is mounted transverse (perpendicular) to the principal axis of strain.
Completion resistors provide half bridge completion.
Sensitive to both axial and bending strain.
Compensates for temperature.
Compensates for the aggregate effect on the principle strain measurement due to the Poisson’s ratio of the specimenmaterial.
Sensitivity at 1000is ~ 0.65 mVout/ VEX input.
The following apply to the above circuit diagram and to the equations provided in the supplemental information.
R1 and R2 are half-bridge completion resistors.
R3 is the active strain-gauge element measuring compression from Poisson effect (– ).
R4 is the active strain-gauge element measuring tensile strain (+ ).
Supplemental Information
Data from strain channels (Half-Bridge I) is converted to strain units using the following equation:
4 Vr

6

10

GF ( 16224 User’s Manual

)

2Vr (

1)

1

RL
Rg

Strain and Pressure/Force

7-5

Half-Bridge II
Half-bridge type II only measures bending strain.

Half-bridge type II has the following characteristics:
Two active strain-gauge elements. One is mounted in the direction of bending strain on one side of the strain
specimen (top), the other is mounted in the direction of bending strain on the opposite side (bottom).
Completion resistors provide half bridge completion.
Sensitive to bending strain.
Rejects axial strain.
Compensates for temperature.
Sensitivity at 1000is ~ 1 mVout/ VEX input.
The following symbols apply to the above circuit diagram and to the equations provided in the supplemental information.
R1 and R2 are half-bridge completion resistors.
R3 is the active strain-gauge element measuring compressive strain (– ).
R4 is the active strain-gauge element measuring tensile strain (+ ).

Supplemental Information
Data from strain channels (Half-Bridge II) is converted to strain units using the following equation:
6

10

2 Vr
GF

1

RL
Rg

7-6 Strain and Pressure/Force

6224 User’s ManualFull-Bridge I
The Full-bridge type I only measures bending strain.

Full-bridge type I configuration has the following characteristics:
Four active strain-gauge elements. Two are mounted in the direction of bending strain on one side of the strain
specimen (top), the other two are mounted in the direction of bending strain on the opposite side (bottom).
Highly sensitive to bending strain.
Rejects axial strain.
Compensates for temperature.
Compensates for lead resistance.
Sensitivity at 1000is ~ 2.0 mVout/ VEX input.
The following symbols apply to the above circuit diagram and to the equations provided in the supplemental information.
R1 is an active strain-gauge element measuring compressive strain (– ).
R2 is an active strain-gauge element measuring tensile strain (+ ).
R3 is an active strain-gauge element measuring compressive strain (– ).
R4 is an active strain-gauge element measuring tensile strain (+ ).
Supplemental Information
Data from strain channels (Full-Bridge I) is converted to strain units using the following equation:

6

10

Vr
GF

6224 User’s Manual

Strain and Pressure/Force

7-7

Full-Bridge II
Full-bridge type II only measures bending strain.

Full-bridge type II configuration has the following characteristics:
Four active strain-gauge elements. Two are mounted in the direction of bending strain with one on one side of thestrain specimen (top), the other on the opposite side (bottom). The other two act together as a Poisson gauge and aremounted transverse (perpendicular) to the principal axis of strain with one on one side of the strain specimen (top),
the other on the opposite side (bottom).
Rejects axial strain.
Compensates for temperature.
Compensates for the aggregate effect on the principle strain measurement due to the Poisson’s ratio of the specimenmaterial.
Compensates for lead resistance.
Sensitivity at 1000is ~ 1.3 mVout/ VEX input.
The following symbols apply to the above circuit diagram and to the equations provided in the supplemental information.
R1 is an active strain-gauge element measuring compressive Poisson effect (– ).
R2 is an active strain-gauge element measuring tensile Poisson effect (+ ).
R3 is an active strain-gauge element measuring compressive strain (– ).
R4 is an active strain-gauge element measuring tensile strain (+ ).

Supplemental Information
Data from strain channels (Full-Bridge II) is converted to strain units using the following equation:
6

10

2 Vr
GF ( 1)

7-8 Strain and Pressure/Force

6224 User’s ManualFull-Bridge III
Full-bridge type III only measures axial strain.

A Full-bridge type III configuration has the following characteristics:
Four active strain-gauge elements. Two are mounted in the direction of axial strain with one on one side of the strain
specimen (top), the other on the opposite side (bottom). The other two act together as a Poisson gauge and aremounted transverse (perpendicular) to the principal axis of strain with one on one side of the strain specimen (top),
the other on the opposite side (bottom).
Compensates for temperature.
Rejects bending strain.
Compensates for the aggregate effect on the principle strain measurement due to the Poisson’s ratio of the specimenmaterial.
Compensates for lead resistance.
Sensitivity at 1000is ~ 1.3 mVout/ VEX input.
The following symbols apply to the above circuit diagram and to the equations provided in the supplemental information.
R1 is an active strain-gauge element measuring compressive Poisson effect (– ).
R2 is an active strain-gauge element measuring tensile strain (+ ).
R3 is an active strain-gauge element measuring compressive Poisson effect (– ).
R4 is an active strain-gauge element measuring the tensile strain (+ ).

Supplemental Information
Data from strain channels (Full-Bridge III) is converted to strain units using the following equation:
2 Vr

6

10

GF ( 16224 User’s Manual

)

Vr (

1)

Strain and Pressure/Force

7-9

Pressure/Force
Pressure is defined as force per unit area that a fluid exerts on its surroundings. For example, pressure, P, is afunction of force, F, and area, A.
P = F/A

Data from pressure/force channels is converted using the following equation:

P

O fs

Vm

S

Vex

Where:
P = Pressure
S = the sensitivity of the gauge in mV/V
Ofs = the Full Scale Output (Outputfs)
Vm = the measured voltage
Vex = the excitation voltage

7-10 Strain and Pressure/Force

6224 User’s ManualProduct Care and Customer Support

8

Electrostatic Discharge (ESD) Handling Notice…… 8-1
Product Care …… 8-1
The ReadMe File and the Install CD-ROM …… 8-2
Customer Support …… 8-2

Electrostatic Discharge (ESD) Handling Notice
The discharge of static electricity can damage some electronic components. Semiconductor devices areespecially susceptible to ESD damage. You should always handle components carefully, and you should
never touch connector pins or circuit components unless you are following ESD guidelines in an
appropriate ESD-controlled area. Such guidelines include the use of properly grounded mats and wrist
straps, ESD bags and cartons, and related procedures.

Product Care
6000 series devices are essentially maintenance free and require only a minimal amount of care. They should betreated much like any other high-tech piece of equipment. In general:
Operate the units in ventilated and relatively dust-free environments.
Keep the units clear of harsh chemicals and abrasive elements.
Avoid exposing the products to extreme heat; for example, avoid setting the units near boilers andfurnaces.
Avoid extreme shock and vibration.
Avoid subjecting the units to liquids and extremely fine air particulate, such as silica dust.
Never open the unit. The unit should only be opened by qualified service technicians.
A “common-sense” approach to handling acquisition components will go a long way in protecting them frominadvertent damage.

Note that you can use lint-free rags and Isopropyl Alcohol (Rubbing Alcohol) to clean the outer surfaces of a 6000
series device.

6000 Series

Product Care and Customer Support

8-1

The Readme File and the Install CD-ROM
The Install CD-ROM includes a ReadMe file. The ReadMe may contain late-breaking information that does notappear in other documentation. During installation you should review the ReadMe file when prompted to by the
program.
The Install CD-ROM includes:
Windows drivers
Encore, interactive measurement software*
User documentation in Adobe® PDF format

Customer Support
If you need to report problems or request product support . . .
Note: Please do not return any equipment to the factory unless it has an RMA number (Return Merchandise
Authorization number). RMA numbers are issued by the factory.
To report problems or request support, contact our Application Engineering department. Contact information
follows shortly. When you contact us, please have the following information available, as applicable:
Hardware model numbers and software version numbers.
Operating system, type of computer, and device information in the Windows control panel, such as
interrupts and address settings for our hardware and others.
Results of any tests.
Hardware setup and software configuration.
Information on input signals, including voltage ranges, signal impedance ranges, noise content, andcommon mode voltages.
The name of a contact person from your company who can discuss the problems encountered.
Instructions for return shipping.
All troubleshooting notes and comments on tests performed, and all problem-related conditions.
Note: Before calling for assistance, take a few minutes to read all parts of the manual that may be relevant to the
problem. Also, please review the troubleshooting material.
You can reach Measurement Computing by one of the following means:
Phone: (508) 946-5100
Fax:(508) 946-9500
E-mail: info@mccdaq.com
Internet: www.mccdaq.com
Mail:
Measurement Computing  10 Commerce Way

 Norton, MA

02766

All equipment returned to the manufacturer must have an RMA (Return Material Authorization) number. You canobtain an RMA number from our Application Engineering department. When returning the equipment, use theoriginal shipping container (or equivalent) to prevent damage. Put the RMA number on your shipping label toensure that your shipment will be handled properly. After receiving your equipment, we will fax a confirmation
form that summarizes the charges (if applicable) and expected return date.

* Encore includes a Comprehensive Help (.chm), a Getting Started with Encore manual, (.pdf) andQuick Start Video Tutorial (.wmv).

8-2

Product Care and Customer Support

6000 Series

Specifications – 6224

9

The following specifications are typical for the range –40 to 50°C, unless otherwise noted.

Input Characteristics
Number of analog channels ................................................ 12
Bridge completion
Full and half ................................................................ Internal
Quarter......................................................................... External
ADC resolution................................................................... 24 bits
Type of ADC ...................................................................... Delta-sigma (with analog pre-filtering)
Sampling mode ................................................................... Simultaneous
Data rates ( ) ..................................................................... (50kS/s)/= 1, 2, ... 31.
Multiple device, channel sync skew1 .................................. 1 Sample Period
Single device, channel-to-channel
matching (calibrated) .................................................... 350nS (max)Master timebase (internal)
Frequency ........................................................................... 12.8 MHz
Accuracy............................................................................. 100 ppm max
Nominal full-scale range
25 mV/V
Scaling coefficient ............................................................. 2.9802 nV/V per LSB
Overvoltage protection between any two terminals ........... 30 V

Accuracy
Percent of
Reading

OffsetCalibrated, max (–40 to 50 °C)

0.20%

0.0625 mV/V

Calibrated, typ (25 °C, ±5 °C)

0.05%

0.0125 mV/V

Measurement Conditions** Excluding offset null or shunt calibration.
Gain drift………………………………… ......................... 10 ppm/°C max
Offset drift
2.5 V excitation ................................... ....................... 0.6 µV/V per °C
3.3 V excitation ................................... ....................... 0.5 µV/V per °C
5 V excitation ...................................... ....................... 0.3 µV/V per °C
10 V excitation .................................... ....................... 0.2 µV/V per °C

Channel-to-channel matching (calibrated)1
Input Signal Frequency (fin)Gain
TypicalMaximum

0 to 1 kHz

0.15%

0.3%

1 to 20 kHz

0.4%

1.1%

Phase nonlinearity
0 to 1 kHz
0.001°
0 to 20 kHz
0.1°
Input delay ......................................................................... 4.8 µs + 38.4/
Refer to the synchronization section of chapter 1 for information regarding multiple devices.
6224 User’s Manual

Specifications

9-1

Specifications are subject to change without notice.

PassbandFrequency ..................... .............................................. ... 0.45 ·Flatness ........................................................................ 0.1 dB max
StopbandFrequency ................................................................... 0.55 ·Attenuation ................................................................. 100 dB
Alias-free bandwidth .......................................................... 0.45 ·Oversample rate .................................................................. 64 ·Attenuation at oversample rate1
50 kS/s ......................................................................... 90 dB @ 3.2 MHz
10 kS/s ......................................................................... 60 dB @ 640 kHz
Common-mode voltage
All signals to earth ground
60 VDC
Common-mode rejection ratio (CMRR)
Relative to earth ground2 ............................................. -140 dB @ 0 to 60 Hz
Relative to EX– ........................................................... -85 dB @ 0 to 1 kHz

Input noise
Excitation
Voltage
2.5 V
3.3 V
5.0 V
10.0 V

Density(nV/Vrms per
8
6
4
2

Total
)

(50kS/S)(µV/Vrms)

0 to 1 kHz(nV/Vrms)

1.3
1.0
0.6
0.3

250
190
130
65

Spurious-free dynamic range (SFDR) ................................ 106 dB, (1KHz, –60 dBFS)
Total harmonic distortion (fundamental @ –20 dBFS)
1 kHz ........................................................................... -100 dB
8 kHz .......................................................................... -90 dB
Excitation noise .................................................................. 0.1 mV/Vrms
Crosstalk1 kHz ........................................................................... -110 dB
10 kHz ........................................................................ -100 dB
Shunt calibration
Resistance .................................................................... 100 k
Resistor accuracy
25°C
110–40 to 50°C
200
Excitation
Internal voltage ............................................................ 2.5 V, 3.3 V, 5.0 V, 10.0 V
Internal power ............................................................. 450 mW max
External voltage........................................................... 2 V to 10 V

Rejection by analog pre-filter of signal frequencies at oversample rate.
Measured with a balanced cable. Shielded cables may be significantly unbalanced.
9-2

Specifications

6224 User’s ManualSpecifications are subject to change without notice.

Digital I/O
Channels:

8 Digital I/O, programmable as asingle port, or as individual lines

Power-up mode:

Inputs pulled low

Connector:

DB-9 female (figure)

Digital I/O
Channel/Line
Numbers

DB9 – As viewed from the rear panel.
Programmable Input Scanning Modes:
Asynchronous:............................................................. Under program control at any time relative to analog
scanning.
Synchronous: ............................................................... Data captured synchronously with the analog
channels.
Input Levels:
Low: ........................................................................... 0 to +0.8V
High: ........................................................................... +2.0V to +5.0V
Input Voltage Range without Damage: .............................. -0.6V to+5.6V max
Input pull-down resistor: .................................................... 10kΩ
Output voltage range: ........................................................ 0 to +3.0V(may be externally pulled up to 5.6V without damage)
Output resistance: ............................................................... 40Ω
Output Levels: (see 6000 Series Digital Output graph below)
Sampling: .......................................................................... 1 MHz max
continuous

Input / Output Circuit

Output timing: ................................................................... Outputs are always written asynchronously.

6224 User’s Manual

Specifications

9-3

Specifications are subject to change without notice.

Counter Input
Channels ............................................................................. Up to 4 independent
Resolution........................................................................... 32-bit
Input frequency................................................................... 20 MHz maximum
Input characteristics ............................................................ 10 k Ω pull-down
Trigger level ....................................................................... TTL
Minimum pulse width......................................................... 25 ns high, 25 ns low
Programmable modes ......................................................... Counter, Encoder
Encoder resolution .............................................................. x1 (default), x2, and x4
Encoder sources .................................................................. A, B, and Z; can be assigned to any digital pin x.
Counter source .................................................................... Internal Clock, Timer 1, Timer 2, and digital pin x.
One source can be used in multiple counters.
Counter mode options......................................................... Totalize, Clear on Read, Rollover, Stop at the Top,
Increment, Decrement, Rising Edge, Falling Edge
Counter gate options ........................................................... Unused, Internal Clock, Timer 1, Timer 2, andDigital pin x. One gate can be used in multiple
counters.

Power
Power consumption ........................................................... 6.3 W typ, 6.6 W max
Power jack Barrel type: 5.5 mm O.D.;
2.1 mm I.D.

Physical Characteristics
Weight ............................................................................... 1.3 kg (2.88 lb)
Dimensions 276.9mm W x 169.8mm D x 30.5mm H(10.9” W x 6.685” D x 1.2” H)

Environmental
6224 units are intended for indoor use only but may be used outdoors if installed in a suitable enclosure.
Operating temperature ........................................................ -40 to 50°C
Storage temperature ............................................................ -40 to 75°C
Ingress protection ............................................................... IP 40
Operating humidity ............................................................ 10 to 90% RH,
Noncondensing ................................................................. 5 to 95% RH, noncondensing
Maximum altitude .............................................................. 2,000 m (6,562 ft)
Pollution Degree ................................................................. 2

Calibration
Calibration interval ............................................................ 1 year
Contact Measurement Computing for information regarding calibration service.
Note: The above calibration information pertains to hardware calibration, not to be confused with “user” or“software” calibration. When performing a “user” calibration via Encore (or other software) keep inmind that sample rate affects both gain and offset of the hardware, and therefore any “user”
calibration should be performed at the same sample rate that is intended for measurements.

9-4

Specifications

6224 User’s ManualSpecifications are subject to change without notice.

Accessories
TR-60U
CA-1
CA-216CN-268CN-269CN-270CA-272-01
CA-74-1

24VDC @ 0.8A (max) universal power supply
120V IEC AC Mains cable (US)220V IEC Mains Cable (EU)
RJ50, 12 pin Screw Terminal, 4pk
RJ50, 120 ohm Quarter Bridge, 4pk
RJ50, 350 ohm Quarter Bridge, 4pk
RJ50 to RJ50 M/M, 1 meter (3.28ft) Cable, 4pk
RJ12 Shielded Cable, 6-Conductor, Sync, 0.3m (1ft)1

190658A-01

Stacking plate kit; includes:
2 stacking plates
8 screws (8-32x0.5in, Phillips Flat, 82 degree)

HA-210-5-BK:

Handle kit; Includes:
1 molded black plastic handle
2 screws (1/4-20x7/8in, Phillips Pan Head)

CA-192-7C
CA-242
CA-242-7

Ethernet Crossover Cable, 2.133m (7ft) 2,3
Ethernet Patch Cable, 0.457m (1.5ft) 2
Ethernet Patch Cable, 2.133m (7ft) 2

Cable Notes:
1

Up to nine units can be synchronized. The total combined length of the SYNC cables is not toexceed 2.438m (8 ft).

2

Ethernet cable length must be <3 m (9.8 ft) in order for the system to be CE Compliant.

3

Ethernet crossover cables should only be used for direct network connections. In particular,
attempting to connect a device to a Hub using a crossover cable may prevent that network linkfrom functioning. Some modern routers have become an exception by including logic to detect
the crossover cable and allow the network link to function.

6224 User’s Manual

Specifications

9-5