Subminiature, FlushDiaphragm Pressure
Transducers
Model S
150 TO 20,000 psi

1-888-282-9891

FLUSH DIAPHRAGM

Model SHIGH FREQUENCY
Our full line of subminiature pressure transducers accurately measure pressure ranges from150 psi to 20,000 psi. These subminiature pressure transducers have a high natural frequency
and utilize a flush diaphragm.
Temperature compensation is accomplished by using temperature sensitive components locatedinside the transducers. These transducers have a small electrical zero balance circuit board
which is in the lead wire (approximately 1" X .087" thick). This balance board does not have tobe the same temperature as the transducers. All units have four (4) active bonded strain gages
arranged in a Wheatstone bridge configuration.

www.honeywell.com/sensing

Sensotec Sensors

Dimensions

PR-18

Model S
Available Pressure Ranges*
150; 200; 300; 500; 750; 1000; 1500;2000; 3000; 5000; 7500; 10,000; 15,000; 20,000 psi“T”
A"D"
Order CodeThread
Dia.
Dia.
B"G"
Diaphragm
BP357
3/8-24 UNF.310.50.45.69
WeldedBP358
7/16-20 UNF.375.56.50.75
Welded* Stocked units (bold face print) available with 3/8- 24 UNF thread only. 3/8-24 UNF thread is only available on
ranges 15,000 psi. Consult factory for ranges below 150 psi.

Installation
Standard “S” type transducers have straight threads and use an O-ring for pressure sealing. To
get the best seal with the O-ring on the transducer, the tapped hole should have the dimensions
shown below. For normal operating temperatures (-65° F to +250° F) use BUNA-N (black)) Orings. For high temperatures (250° F to 425° F) use silicone ru bber (red).
“T”“A”“B”+.000"“D”“D”-.002"

O-Ring

3/8 - 24 UNF
7/16 - 20 UNF

.47"
.54"

.30"
.36"

.445"
.504"

#11
#12

Max.
Max.Torque
Torque(For 17-4 PH only)

300 In-lbs.
500 In-lbs.

1-888-282-9891
Model SPERFORMANCE

Pressure Ranges..........................

3
7/8-24 UNF Thread: 150 to 15,000 psi (10 bar to 1050 bar)
/16-20 UNF Thread: 150 to 20,000 psi (10 bar to 1375 bar)

Non-Linearity andHysteresis (max) ......................
Non-repeatability (max) ................
Output (standard) .........................
Resolution ....................................

ENVIRONMENTAL

1.0% F.S.
+ /-0.1% F.S.
2mV/V
Infinite

Temperature, Operating ...............
Temperature, Compensated ........
Temperature Effect– Zero (max).............................
– Span (max)............................

.01% F.S./ ° F (0.018% F.S./ ° C)
.02% Rdg./ ° F (0.036% Rdg./ ° C)

ELECTRICAL

Strain Gage Type .........................
Excitation (calibration) ..................
Bridge Resistance ........................
Wiring Code (std) .........................
Electrical Termination (std) ..........

Bonded foil
5VDC
350 ohms#1 (See Pg. AP-7)
4 twisted leads (5 ft.)

MECHANICALMedia............................................
Overload-Safe ..............................
Wetted Parts Material...................
Type (Gage, Abs.) ........................
Case Material ...............................

Gas, Liquid
50% over capacity
17-4 PH StainlessGage
17-4 PH Stainless-65° to 300° F (-54° to 150° C)
60° to 160° F (15° to 70° C)

Premium Options: 1d, 1e, 1g, 1h, 1i
Custom threads, metric threads, underwater cable, microtech connector, special headers and400° F (200° C) temperature are available on some models and ra nges, consult factory.
Accessories: Mating connectors and connector/cable assemblies

General Information

SUBMINIATURE FLUSH DIAPHRAGM

Temperature compensated 1b, 1c

PRESSUREOptions (See Appendix)

How to order (See Pg. AP-19)
Gage/Absolute pressure selection flow chart (See Pg. PR-1)●お問い合わせは下記へ ̶PR-19