MODEL 4610 ACCELEROMETERSPECIFICATIONSMEMS DC Accelerometer
Ultra-Stable, DC to 2000Hz Response
Exceptional Thermal Performance<2.0% Total Error Band
<0.1% Linearity Accuracy
Self-test Function Included.

FEATURES





±2g to ±200g Dynamic Range
Self-test Enabled
Amplified Output, Signal Conditioned
Gas Damped MEMS Sensors
Integral Strain Relief
4 to 30Vdc Excitation Voltage
6000g Shock Protection

APPLICATIONSFlight Testing
Flutter and Nacelle Vibrations
Structural Testing
Test and Instrumentation
Performance Testing
Transportation

SENSOR SOLUTIONS /// Model 4610 Rev G

The Model 4610 is an ultra-stable MEMS DC
accelerometer with exceptional performance over a fulloperating temperature range of -55°C to +125°C. The
accelerometers are available in ranges from ±2 to ±200gwith a wide bandwidth from DC to 2000Hz. The model
4610 accelerometers incorporate gas damped variablecapacitance MEMS sensing element with integral overrange stops for high-g shock protection. The
accelerometers are designed for 4 to 30Vdc excitation
voltage and include a self-test option.
For a triaxial version, TE Connectivity also offers themodel 4630 and 4835A accelerometers.

08/2019

Page 1

MODEL 4610 ACCELEROMETERPERFORMANCE SPECIFICATIONS
All values are typical at +24°C, 80Hz and 12Vdc excitation unless otherwise stated. TE Connectivity reserves the right to update and change these
specifications without notice.
Parameters
DYNAMIC
Range (g)Sensitivity, Differential (mV/g)
Frequency Response (Hz)Frequency Response (Hz)Non-Linearity (%FSO)
Transverse Sensitivity (%)
Damping Ratio
Shock Limit (g)Residual Noise (µV RMS)
Spectral Noise (µg/Hz)±210000-2500-500±0.1
<30.7
6000
36014ELECTRICAL
Zero Acceleration Output (mV)
Excitation Voltage (Vdc)
Excitation Current (mA)Common Mode Voltage (Vdc)
Full Scale Output (differential)
Full Scale Output (single-ended)
Output Resistance (Ω)
Insulation Resistance (MΩ)
Turn On Time (msec)
Ground Isolation±50
4 to 30<71.22±2 Vpk (FSO=2V)+0.22 to 2.22 Vpk (FSO=1V)<100>100<100Isolated from Mounting Surface

Differential

ENVIRONMENTAL
Thermal Zero Shift (%FSO/°C)
Thermal Sensitivity Shift (%/°C)
Operating Temperature (°C)
Storage Temperature (°C)
Humidity (MEMS Sensor and Electronics)
Humidity (Housing)±0.004±0.008-55 to 125-55 to 125
Hermetically Sealed
Epoxy Sealed, IP65

TypicalTypicalPHYSICAL
Case Material
Cable
Weight (grams)
Mounting
Mounting TorqueAnodized Aluminum
5x #30 AWG Conductors PFA Insulated, Braided Shield, TPE Jacket
8
2x #4 or M3 Screws
6 lb-in (0.7 N-m)

Notes
±5400
0-700
0-1000±0.1
<30.7
6000
38028±102000-1000
0-1500±0.1<30.76000
400
45±30670-15000-2000±0.1<30.76000
440
137±50400-1500
0-2000±0.1<30.76000
480
231±100200-1500
0-2000±0.1<30.76000
500
464±200
100-1500
0-2000±0.1<30.76000
500920±5%±5%±1dB<1 Typical
Passband
Passband@100Vdc

Calibration supplied:

CS-FREQ-0100

NIST Traceable Amplitude Calibration from 20Hz to ±5% Frequency Response Limit

Supplied accessories:

AC-A02285

2x #4-40 (7/16 inch length) Socket Head Cap Screw and Washer

Optional accessories:

AC-D02669
AC-D02744
121

Triaxial Mounting Block
Adhesive Mounting Adaptor
Three Channel DC Differential Amplifier

The information in this sheet has been carefully reviewed and is believed to be accurate; however, no responsibility is assumed for inaccuracies. Furthermore, thisinformation does not convey to the purchaser of such devices any license under the patent rights to the manufacturer. TE Connectivity reserves the right to makechanges without further notice to any product herein. TE Connectivity makes no warranty, representation or guarantee regarding the suitability of its product for any
particular purpose, nor does TE Connectivity assume any liability arising out of the application or use of any product or circuit and specifically disclaims any and all
liability, including without limitation consequential or incidental damages. Typical parameters can and do vary in different applications. All operating parameters must
be validated for each customer application by customer’s technical experts. TE Connectivity does not convey any license under its patent rights nor the rights ofothers.

SENSOR SOLUTIONS /// Model 4610 Rev G

08/2019

Page 2

MODEL 4610 ACCELEROMETERDIMENSIONSSCHEMATICSENSOR SOLUTIONS /// Model 4610 Rev G08/2019

Page 3

MODEL 4610 ACCELEROMETER

ORDERING INFORMATION
4610

GGG

ZZZ

DRange002 = 2g
005 = 5g
010 = 10g
030 = 30g
050 = 50g
100 = 100g
200 = 200g
Cable length
120 = 120 inches, 10 feet
240 = 240 inches, 20 feet
360 = 360 inches, 30 feet
480 = 480 inches, 40 feet
600 = 600 inches, 50 feet
197 = 197 inches, 5 meters
394 = 394 inches, 10 meters
Example;4610-010-120-D
Model 4610, 10g range, 120inch (10ft) cable length

MEAS France SAS and Measurement Specialties (China) , Inc., are TE Connectivity companies.
TE Connectivity, TE, TE connectivity (logo) are trademarks. All other logos, products and/or company names referred to herein might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes nowarranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard Terms
and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuse of theproduct. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2017

TE Connectivity Ltd. All Rights Reserved.

SENSOR SOLUTIONS /// Model 4610 Rev G

08/2019

Page 4