Sensepoint

The Sensepoint range of flammable,
toxic and Oxygen gas detectors offer
users a high quality, low cost solution
to their industrial gas monitoring needs.

Sensepoint

Fit For Purpose•	Flammable, toxic and Oxygen
versions available• High performance, low cost
• Suitable for new and retro fit applications

Proven Sensor Technology••••

SurecellTM electrochemical sensors
Poison resistant catalytic bead
Fast and reliable
Typical 2 to 5 years operational life

ATEX/IECEx Certified•	Complies with the latest standards• Hazardous area approval and performance• IP65/67 for use in harsh environments

Easy to Use•••Factory set operation
Simple to replace
Minimal training required

Range of Accessories••••

Weatherproof cap
Flow adaptor
Duct mounting kitCollecting cone

The Sensepoint range of flammable, toxic
and Oxygen gas detectors offer users a highquality, low cost solution to their industrial gas
monitoring needs. Installation in potentiallyexplosive atmospheres is made by the use ofa suitable Ex d or Ex e approved junction box.
The industry standard outputs fromSensepoint are connected to either a choiceof Honeywell Analytics’ controllers or 3rdparty DCS/ PLC systems. Both flammable andtoxic versions are certified to the latest ATEX/
IECEx safety and performance standards.

The flammable detector is available in either % LEL
or PPM detection range versions and provides
an industry standard mV bridge output.
Typical life expectancy is greater than 5 years,
after which the detector is simply replaced with anew one.
The toxic detectors are supplied pre calibrated foreasy installation and provide an industry standard
2 wire 4-20mA loop output. User connections
are Ex d, while front end Intrinsic Safety (IS) design
allows for easy exchange of the replaceable
sensor while powered. IS design also removes
the need for a sinter enabling very fast speedof response and detection of ‘sticky’ gases suchas Chlorine and Ammonia.
Sensepoint has proven its suitability across a wide
range of demanding applications by delivering highperformance and reliability at an affordable cost.

Installation Details

Dimensions
Flammable % LEL or PPM Version

Toxic or Oxygen Version
100mm

100mm

Standard
Junction Box

100mm

100mm

Standard
Junction Box

Ex e/Ex d glandEx e/Ex d glandThreads available: M20,
M25 and ¾ NPT

Threads available: M20,
M25, M26, ¾ NPT(ppm M20 and ¾ NPT)

Serial Number/year of
Manufacture

90mm

74mm

Serial Number/year of
Manufacture

Sensepoint Toxic
Detector

Sensepoint Flammable %LELor ppm Detector
FilterFilterFilter Housing

52mm(across flats)

56mm

56mm

Filter Housing

52mm(across flats)

Sensepoint flammable and toxic gas detectors are installed using suitable Ex d or Ex e junction boxes.
The standard junction box available from Honeywell Analytics includes an earth continuity plate foradded RFI protection. The detectors have a choice of mounting thread size to match local preferences.
The Sensepoint range of flammable and toxic detectors are designed for use with a wide rangeof controllers available from Honeywell Analytics or with 3rd party control equipment accepting
their industry standard output signals.

Installation Details

Electrical Connections

Typical Terminal Connections

Flammable % LEL Version+VE

Blue (NS)-VE
SignalBrown (S)To Controller

White (01)

3 Wire mV Bridge Circuit

Toxic or Oxygen Version+VEBlue (-)

SignalCable ScreenTo Controller

Red (+)
Screen2 Wire 4-20mA (Sink)

Flammable PPM Version+VE

Blue (NS)-VE
SignalGreen (01)Red (S)To Controller
Cable Screen

Screen3 Wire mV Bridge Circuit

Technical Specification

Gas & Range

Flammable
0-100% LEL

Flammable
HydrogenCarbon
PPM (Typically Sulphide (H2S) Monoxide (CO)
0-10% LEL) 20/50/100ppm 100/200/500ppm

Ammonia(NH3)50/1000ppm

Chlorine(Cl2)
5/15ppm

Operating Temp

-40 to +80˚C1-30 to +65˚C-20 to +40˚C-20 to +50˚C-25 to +50˚C

Operating Humidity
Storage Conditions-20 to +50˚C

HydrogenSulfur Dioxide
(H2)
(SO2)
1000/10000ppm
15/50ppm

Nitric Oxide(NO)
100ppmNitrogenDioxide (NO2)
10ppm

12 hours

1 hour

-15 to +40˚C

Continuous: 20 to 90% RH (non-condensing). Intermittent: 10 to 99% RH (non-condensing)
0 to 25˚C; 30 to 70% RH

0 to 25˚C; 30 to 70% RH

Operating PressureWarm-up TimeOxygen(O2)
25% V / V

90 to 110kPa< 10 minutes< 20 minutes< 3 minutes< 5 minutes< 3 minutes

Response TimeT20< 6 secs< 5 secs–< 10 secs< 3 secs––< 3 secs

< 15 secs< 5 secs< 15 secs< 12 secs< 10 secs< 20 secs< 5 secs< 10 secs< 10 secs< 5 secs< 30 secs

T90< 10 secs< 40 secs< 30 secs< 65 secs< 105 secs

< 10 secs< 45 secs< 90 secs< 30 secs< 60 secs

Supply200mA; 0.7W maximum

18 to 30VDC; 0.9W maximum

mV bridge

Loop powered 4-20mA< 5% fsd or ±10% of reading< 5% fsd

T50Signal Output
Linearity
Operating Life

5 years (typical)

Storage Life

Typically, over 5 years when inoriginal sealed packaging

2 years (typical)

18 months (typical)

1 year (typical)

18 months (typical)

6 months.

Stability (zero)with time*with temperature**
with humidity< ±3% fsd / yr
< ±3% fsd

< ±1% fsd2

< ±2% fsd / yr
< ±5% fsd

< ±1% fsd

< ±10% fsd< ±2% fsd

< ±3% fsd / yr

< ±2% fsd / yr

< ±4% fsd / yr

< ±25% fsd< ±20% fsd< ±10% fsd< ±2% fsd

< ±6% fsd

< ±2% fsd

< ±2% fsd

< ±5% fsd

< ±10% fsd< ±2% fsd

< ±2% fsd

< ±10% fsd

Stability (span)with time*with temperature***with humidity
Weight< ±3% fsd

< ±8% fsd

< ±2% fsd
190g

< ±20% fsd / yr < 10% fsd / yr < ±20% fsd / yr
< 40% fsd

< ±10% fsd< ±60% fsd< ±10% fsd< ±2% fsd

< ±1% fsd

< ±2% fsd

185g

205g

Construc. Materials

Polyphenylene Sulphide (PPS)

I.P. (NEMA) Rating

IP65 (standard), IP67 (with weather protection fitted)

Certification Details(ATEX/ IECEx)

Other Certifications
ApplicableStandards

II 2 GD Ex d IIC Gb T85ºC
Tamb -40ºC to +55ºC
II 2 GD Ex d IIC Gb T100ºC
Tamb -40ºC to +70ºC
II 2 GD Ex d IIC Gb T135ºC
Tamb -40ºC to +80ºC
tb IIIC A21 Db IP67
Baseefa08ATEX0265X
IECEx BAS08.0068X< ±2% fsd / yr

< ±20% fsd< ±15% fsd< ±25% fsd< ±10% fsd
185g

II 2 GD Ex d ia IIC T4 Gb Tamb -40ºC to +65ºC
tb IIIC A21 IP67 T135ºC Db
Baseefa08ATEX0263X
IECEx BAS08.0070X

Russian: GOST
This product complies with the relevant CE standards including: EMC to EN50270, Flammable sensor to EN60079-29-1

NOTES
1

High temp range version available -55 to +150˚C – See separate datasheet DS0303

2

Over temp range -20 to +55˚C* = % of range (fsd) per year** Over operating temperature range*** = % of applied concentrationOrdering Information
Accessories

Sensepoint Combustible

00780-A-0100	Term Hsg – Bartec DE1155 With Continuity Plate
1 x 25mm – 3 x 20mm Entries – ATEX Approved
00780-A-0160 Right Angle Mounting Bracket For Ceiling Mounting
02000-A-1635 Series 2000 Weather Housing Use With Toxic Sensor
02000-A-1640	Series 2000 Weather Housing use With Flammable Sensor
02000-A-1642	Series 2000 Collecting Cone Use With Flammable Sensor
02000-A-1645	Series 2000 Calibration Flow Housing Assembly (Flam and Toxic)
02000-A-3120 Series 2000 Flow Housing Assembly Plastic
2106B1110
Catalytic Bridge To 4-20mA Converter
2106B2114	Toxic Calibration Kit including Calibration Cover,
Removal Tool, Potentiometer Adjustment Tool
2106D2055
M36 / M40 Thread Adaptor
225-262-138
Cable Gland And Caps, 20mm, Nylon

2106B1200
2106B1201
2106B1202
2106B1203
2106B1204
2106B1205
2106B1209

2106B1545
2106B1546
2106B1547
2106B1548
2106B1549
2106B1594
2106B1595
2106B1596
2106B1597
2106B1598
2106B1599

O2 Electrochemical Cell Exchange Kit
SO2 Electrochemical Cell Exchange Kit
Cl2 Electrochemical Cell Exchange Kit
CO Electrochemical Cell Exchange Kit
H2S Electrochemical Cell Exchange Kit
NO Electrochemical Cell Exchange Kit
NH3 0-1000ppm Electrochemical Cell Exchange Kit
NH3 0-50ppm Electrochemical Cell Exchange Kit
H2 0-1000ppm Electrochemical Cell Exchange Kit
H2 0-10000ppm Electrochemical Exchange Kit
NO2 0-10ppm Electrochemical Exchange Kit

Calibration Equipment			Contact your local Honeywell Analytics representativefor details of suitable calibration equipment

Find out more
www.honeywellanalytics.com
Contact Honeywell Analytics:
Europe, Middle East, Africa, India
Life Safety Distribution AG
Javastrasse 28604 Hegnau
Switzerland
Tel: +41 (0)44 943 4300
Fax: +41 (0)44 943 4398
India Tel: +91 124 4752700
gasdetection@honeywell.com
Americas
Honeywell Analytics Inc.
405 Barclay Blvd.
Lincolnshire, IL 60069
USATel: +1 847 955 8200
Toll free: +1 800 538 0363
Fax: +1 847 955 8210
detectgas@honeywell.com
Asia Pacific
Honeywell Analytics Asia Pacific#508, Kolon Science Valley (I)187-10 Guro-Dong, Guro-Gu
Seoul, 152-050
Korea
Tel: +82 (0)2 6909 0300
Fax: +82 (0)2 2025 0329
analytics.ap@honeywell.com
Technical Services
EMEAI: HAexpert@honeywell.com
US:
ha.us.service@honeywell.com
AP:
ha.ap.service@honeywell.com
www.honeywell.com

Sensepoint Toxic
2106B1400
2106B1401
2106B1402
2106B1500
2106B1501
2106B1502
2106B1700
2106B1701
2106B1702
2106B1405
2106B1406
2106B1407
2106B15052106B1506
2106B1507
2106B1705
2106B1706
2106B1707
2106B1410
2106B1411
2106B1510
2106B1511
2106B1711
2106B1413
2106B1414
2106B1415
2106B1513
2106B1514
2106B1515
2106B1713
2106B1714
2106B1715
2106B1416
2106B1417
2106B1516
2106B1517
2106B1716
2106B1717
2106B1418
2106B1518
2106B1718
2106B1420
2106B1421
2106B1520
2106B1521
2106B1720
2106B1721
2106B1422
2106B1522
2106B1722
2106B1430
2106B1530
2106B1730

Please Note:
While every effort has been made to ensure
accuracy in this publication, no responsibility
can be accepted for errors or omissions.
Data may change, as well as legislation, and you
are strongly advised to obtain copies of the mostrecently issued regulations, standards, and guidelines.
This publication is not intended to form thebasis of a contract.

H_Sensepoint_DS01069_V6_EMEAI
09/12© 2012 Honeywell Analytics

Sensepoint H2S 0-20 ppm Toxic Sensor M20
Sensepoint H2S 0-50 ppm Toxic Sensor M20
Sensepoint H2S 0-100 ppm Toxic Sensor M20
Sensepoint H2S 0-20 ppm Toxic Sensor M25
Sensepoint H2S 0-50 ppm Toxic Sensor M25Sensepoint H2S 0-100 ppm Toxic Sensor M25
Sensepoint H2S 0-20 ppm Toxic Sensor ¾”NPTSensepoint H2S 0-50 ppm Toxic Sensor ¾”NPTSensepoint H2S 0-100 ppm Toxic Sensor ¾”NPT
Sensepoint CO 0-100 ppm Toxic Sensor M20
Sensepoint CO 0-200 ppm Toxic Sensor M20
Sensepoint CO 0-500 ppm Toxic Sensor M20
Sensepoint CO 0-100 ppm Toxic Sensor M25
Sensepoint CO 0-200 ppm Toxic Sensor M25
Sensepoint CO 0-500 ppm Toxic Sensor M25
Sensepoint CO 0-100 ppm Toxic Sensor ¾”NPTSensepoint CO 0-200 ppm Toxic Sensor ¾”NPTSensepoint CO 0-500 ppm Toxic Sensor ¾”NPTSensepoint Cl2 0-5 ppm Toxic Sensor M20
Sensepoint Cl2 0-15 ppm Toxic Sensor M20
Sensepoint Cl2 0-5 ppm Toxic Sensor M25
Sensepoint Cl2 0-15 ppm Toxic Sensor M25
Sensepoint Cl2 0-15 ppm Toxic Sensor ¾”NPTSensepoint NH3 0-50 ppm Toxic Sensor M20
Sensepoint NH3 0-1000 ppm Toxic Sensor M20
Sensepoint NH3 0-100 ppm Toxic Sensor M20Sensepoint NH3 0-50 ppm Toxic Sensor M25
Sensepoint NH3 0-1000 ppm Toxic Sensor M25
Sensepoint NH3 0-100 ppm Toxic Sensor M25
Sensepoint NH3 0-50 ppm Toxic Sensor ¾”NPTSensepoint NH3 0-1000 ppm Toxic Sensor ¾”NPTSensepoint NH3 0-100 ppm Toxic Sensor ¾”NPTSensepoint H2 0-1000 ppm Toxic Sensor M20
Sensepoint H2 0-10000 ppm Toxic Sensor M20
Sensepoint H2 0-1000 ppm Toxic Sensor M25
Sensepoint H2 0-10000 ppm Toxic Sensor M25
Sensepoint H2 0-1000 ppm Toxic Sensor ¾”NPTSensepoint H2 0-10000 ppm Toxic Sensor ¾”NPTSensepoint NO 0-100 ppm Toxic Sensor M20
Sensepoint NO 0-100 ppm Toxic Sensor M25
Sensepoint NO 0-100 ppm Toxic Sensor ¾”NPTSensepoint SO2 0-15 ppm Toxic Sensor M20
Sensepoint SO2 0-50 ppm Toxic Sensor M20
Sensepoint SO2 0-15 ppm Toxic Sensor M25
Sensepoint SO2 0-50 ppm Toxic Sensor M25
Sensepoint SO2 0-15 ppm Toxic Sensor ¾”NPTSensepoint SO2 0-50 ppm Toxic Sensor ¾”NPTSensepoint NO2 0-10 ppm Toxic Sensor M20
Sensepoint NO2 0-10 ppm Toxic Sensor M25
Sensepoint NO2 0-10 ppm Toxic Sensor ¾”NPTSensepoint O2 25% V/V Toxic Sensor M20
Sensepoint O2 25% V/V Toxic Sensor M25
Sensepoint O2 25% V/V Toxic Sensor ¾”NPT11577

SparesSensepoint 0-100% LEL Flammable Sensor M20
Sensepoint 0-100% LEL Flammable Sensor M25
Sensepoint 0-100% LEL Flammable Sensor M26
Sensepoint 0-100% LEL Flammable Sensor M26 (VQ1AB pellistor)
Sensepoint 0-100% LEL Flammable Sensor ¾”NPT
Sensepoint ppm (0-10% LEL) Flammable Sensor M20
Sensepoint ppm (0-10% LEL) Flammable Sensor ¾”NPT