ELHM
Miniature Load Cell
SPECIFICATIONS Compression or tension/compression

 Ranges from 2.5 kN to 50 kN (500 lbf to 10 klbf)

 Metal foil strain gages for high stability

The ELHM load cell is a compact package able to fit into many
applications where others cannot.
When compact design and superior stability arerequired, the ELHM load cell is the sensor for your application.
The ELHM is provided with either SAE or metric threads fortension and compression applications.

FEATURES

Care should be exercised to isolate your ELHM from offaxis loads. The ELHM is a Poisson column design providing low
full scale input deflection and superior stability for longer termmeasurements. Designed specifically to provide high zerostability, the ELFM is rated for a cycle life expectancy of typically1 x 106 0-FS cycles of zero to full rated load.

 Low mass compression style ortension compression with studs
 Sensitivity FSO: 2mV/V
 Low noise High reliability

The ELHM can be configured with a variety of different
options to fine tune the instrument to your application: selectfrom several standard compensated temperature ranges, input
voltages, lead lengths or specify entirely unique combinations ofthese options.

APPLICATIONS Theatrical rigging loads
 Assembly forces Weighing
 Tool forces Thrust measurements
 Long period force measurements
 Product validation testing Hoist and winch loads

SENSOR SOLUTIONS /// ELHM_en

On request, Instruction documents can be provided toease the selection and use of our sensors and provide helpful
tips.

06/2017

Page 1ELHM
Miniature Load Cell

STANDARD RANGES
RangesBody style
Overrange

Sensitivity "FSO"

CompressionTension & Compression(nom.)

1.5 x FS

ELHM-B2-

2 mV/V1K

1.5 x FS

ELHM-B2-

2 mV/V

10K

2K

1.5 x FS

ELHM-B3

ELHM-T3

2 mV/V

25K

5K

1.5 x FS

ELHM-B4

ELHM-T4

2 mV/V

50K

10K

1.5 x FS

ELHM-B4

ELHM-T4

2 mV/V

N

lbf

2.5K

500

5K

PERFORMANCE SPECIFICATIONS (typical values at temperature 23±3°C)
PARAMETERSVALUES

NOTES

Sensitivity “FSO”

2mV/V

Supply voltage

5Vdc

Input resistance350Ω nom.

Output resistance350Ω nom.

Non-repeatability± 0 . 2 5 % FSO

Non-linearity±0.3%FS

Hysteresis±0.25%FS

Thermal Zero Shift "TZS"±0.02%FS /°C

Thermal Sensitivity Shift "TSS"±0.02% of reading /°C

Operating temperature-50°C to 120°C

Compensated temperature

20°C to 80°C

Zero offset± 0 . 0 4 mV/V

See option table for other temperatures

Note:
1. Electrical Termination: 3m shielded cable output as standard
2. Type B units have positive output in compression. Type T units have positive output in tension.
3. CE conformance according to EN 61010-1, EN 50081-1, EN 50082-1

SENSOR SOLUTIONS /// ELHM_en06/2017

Page 2

ELHM
Miniature Load CellDIMENSIONS (in metric and imperial)
ELHM-B3 & -B4

ELHM-B2

ITEM

ELHM-B310kN(2 klbf)

ELHM-B4
25 to 50 kN(5 to 10klbf)

A

3m (10 ft)

3m (10 ft)

Bø 10 (0.39”)

ø 18 (0.71”)

C

3 (0.12”)3 (0.12”)Dø 3 (0.12”)

ø 3 (0.12”)

Eø 19 (0.75”)

ø 25 (0.98”)

F

20 (0.79“)25 (0.98”)

G

1 (0.04“)

1 (0.04“)

ELHM-T3 & -T4

INSTALLATIONSENSOR SOLUTIONS /// ELHM_en

ITEM

ELHM-T310 kN(2 klbf)

ELHM-T4
25 to 50 kN(5 to 10 klbf)

A-T3E : 3/8-24 UNF-T3M : M10x1.5-T4E : 5/8-18 UNF-T4M : M16x2

Bø 3 (0.12”)

ø 3 (0.12”)C3m (10 ft)

3m (10 ft)

D

3 (0.12”)3 (0.12”)Eø 19 (0.75”)

ø 25 (0.98”)

F

15 (0.59“)25 (0.98”)

G

20 (0.79“)25 (0.98”)

H

15 (0.59“)25 (0.98”)

CONNECTIONS

06/2017

Page 3

ELHM
Miniature Load CellOPTIONS AND ACCESSORIESOPTIONSCODES

DESCRIPTIONS

Compensated temperature ranges

Z0Z1Z2
Z*-40°C to 20°C
-20°C to 40°C
0°C to 60°C
Non-standard, contact factory

Supply voltage

V00

Replace “00” with voltage between 1 and 10.

Special cable length

L00F
L00M

Replace “00” with total length in feet with Lb range.
Replace “00” with total length in meters with N range.

Connector wired to cable

CRS

Microtech type male or equivalent (w/o mate)
RJ telephone type male (w/o mate)

ORDERING INFORMATION
Model-BodyThread Type-Range & Unit (1)-/Options

ELHM-B2
B2
B3/T3
B4/T4

M : Metric
E : SAE-2.5KN
5KN
10KN
25KN
50KN

-/Z0, Z1, Z2, or Z*/V1 thru V10 or V*/L00F or L00M
/C or RS

(1)Note :500L
1KL
2KL
5KL
10KL

L=lbf ; N=Newton.
Metric threaded units must have range in Newton and cable length in meters specified.
SAE threaded units must have range in lbf and cable length in inches specified.
Options can be multiple if compatible (e.g. ELHM-T3E-2KL-/Z1/L10F/C)

TE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred to herein
might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s StandardTerms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuseof the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies

SENSOR SOLUTIONS /// ELHM_en

All Rights Reserved.

06/2017

Page 4