MODEL 8021-01 & 8022-01
ACCELEROMETERSPECIFICATIONSdimensions

EMI Shielded Industrial Accelerometer
12kHz Bandwidth.
-55°C to +125°C Operating Range.
Side Exit Connector or Integral Cable

The Model 8021-01 & 8022-01 are internally shieldedrugged IEPE accelerometers designed for harshenvironments. The accelerometers are available in ±10g,±80g & ±500g dynamic ranges and have a flat bandwidth
up to greater than 12kHz. The model 8021-01 features aside exit MIL-C-5015 connector while the model 8022-01
features an integral cable. The units are hermetically sealedand are designed to operating in temperature ranges from-55°C to +125°C.

FEATURES





±10g, ±80g & ±500g Dynamic Ranges
Wide bandwidth up to 12kHz
Case Isolated, Internally Shielded
Hermetically Sealed, Welded
Annular Shear Mode
Reverse Wiring Protection
Stable Temperature Response

APPLICATIONS






SENSOR SOLUTIONS /// Model 802X-01 Rev B

General Purpose
Machine Monitoring
Industrial Applications
Harsh Environments
Gearbox Monitoring

9/2015

Page 1

MODEL 8021-01 & 8022-01 ACCELEROMETERPERFORMANCE SPECIFICATIONS
All values are typical at +24°C, 80Hz and 4mA excitation unless otherwise stated. Measurement Specialties reserves the right to update
and change these specifications without notice.
Parameters
DYNAMIC
Range (g)Sensitivity (mV/g)
Frequency Response (Hz)Frequency Response (Hz) 2
Natural Frequency (Hz)Non-Linearity (%FSO)
Transverse Sensitivity (%)
Shock Limit (g)Residual Noise (g RMS)

±105001-40000.5-6000
28000±1<550000.00005

ELECTRICAL
Compliance Voltage (Vdc)
Excitation Current (mA)
Bias Voltage (Vdc)
Output Impedance (Ω)
Insulation Resistance (MΩ)
Warm-up Time (sec)Grounding

18 to 30
2 to 108 to 12<100>100<1Case Isolated, Internally Shielded

ENVIRONMENTAL
Temperature Response (%)
Operating Temperature (°C)
Storage Temperature (°C)

±5-55 to +125-55 to +125

PHYSICAL
Sensing Element
Case Material
Weight (grams)
Mounting TorqueCeramic (shear mode)
Stainless Steel105
24 lb-in (2.7 N-m)

Notes
±80
100
1-80000.5-12000
32000±1<55000
0.0001±500
101-10000
0.5-15000
42000±1<55000
0.0003±10%±10%±3dB

See Note 1
Room Temperature@100Vdc

1For full dynamic range a minimum compliance voltage of 24Vdc is recommended

2

LF option offers a low frequency response to 0.1Hz.

3

Contact factory for Hazardous Area Approval options

Calibration supplied:

CS-FREQ-0100

NIST Traceable Amplitude Calibration from 20Hz to ±10% Frequency Response Limit

Supplied accessories:

AC-A03735¼-28 captive mounting screw

Optional accessories:

316A-XXX
318A-XXX

Cable Assembly, Straight (XXX designates length in inches, 10ft standard)
Cable Assembly, 90 Degrees (XXX designates length in inches, 10ft standard)

The information in this sheet has been carefully reviewed and is believed to be accurate; however, no responsibility is assumed for inaccuracies. Furthermore, thisinformation does not convey to the purchaser of such devices any license under the patent rights to the manufacturer. Measurement Specialties, Inc. reserves the right tomake changes without further notice to any product herein. Measurement Specialties, Inc. makes no warranty, representation or guarantee regarding the suitability of its
product for any particular purpose, nor does Measurement Specialties, Inc. assume any liability arising out of the application or use of any product or circuit and specificallydisclaims any and all liability, including without limitation consequential or incidental damages. Typical parameters can and do vary in different applications. All operating
parameters must be validated for each customer application by customer’s technical experts. Measurement Specialties, Inc. does not convey any license under its patentrights nor the rights of others.

SENSOR SOLUTIONS /// Model 802X-01 Rev B

9/2015

Page 2

MODEL 8021-01 & 8022-01 ACCELEROMETER

ORDERING INFORMATIONPART NUMBERING

Model Number+Range

802XLF-01-GGG-CCC
l lll___Cable length (120 is 120 inches)
l ll_____ Range (080 is 80g)
l l__________ Low frequency option, otherwise leave blankl___________ Electrical Connection Option (1 for 2-pin connector, 2 for integral cable)
Example: 8021-01-080-120
Model 8021-01, 2-pin connector, 80g range, 10ft (120inch) cable length

TE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred to herein
might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard
Terms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuseof the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies

All Rights Reserved.

SENSOR SOLUTIONS /// Model 802X-01 Rev B

9/2015

Page 3