MODEL 832 ACCELEROMETERSPECIFICATIONS

Triaxial Piezoelectric Accelerometer<4µA Current Consumption
Full Signal and Power Conditioning
Circuit Board Mountable

The Model 832 is a low cost, board mountable
triaxial accelerometer. Featuring stable piezo-ceramic
crystals, the accelerometer incorporates full power andsignal conditioning with a maximum current consumptionof 4 micro-amps. The model 832 is available in ±25g to±500g ranges and provides a flat frequency response upto 2kHz. The model 832M1 provides an extended
frequency range to 6kHz.

FEATURES
±25g to ±500g Dynamic Range
Low Cost Triaxial
Hermetically Sealed
Piezo-ceramic Crystals-20° to +80°C Operating Range-40° to +125°C Available on 832M1
Single Axis Configurations Available

APPLICATIONS




SENSOR SOLUTIONS /// Model 832 Rev H

Asset Monitoring
Data Loggers
Impact Monitoring
Machine Health Monitoring
System Wake-Up Switch
Embedded Applications

12/2017

Page 1

MODEL 832 ACCELEROMETERPERFORMANCE SPECIFICATIONS
All values are typical at +24°C, 80Hz and 3.3Vdc excitation unless otherwise stated. TE Connectivity reserves the right to update andchange these specifications without notice.
Parameters
DYNAMIC
Range (g)Sensitivity (mV/g)
Frequency Response (Hz) 1
Natural Frequency (Hz)Non-Linearity (%FSO)
Transverse Sensitivity (%)
Shock Limit (g)Broadband Noise (µV)
Spectral Noise (µg/√Hz)
Spectral Noise (µg/√Hz)
Spectral Noise (µg/√Hz)±2550.02-2000>10000±2<105000
300
12080
40

ELECTRICAL
Bias Voltage (Vdc)
Total Supply Current (µA)
Excitation Voltage (Vdc) 3
Output Impedance (Ω)
Insulation Resistance (MΩ)
Warm-Up Time (msec)
ShieldingGround Isolation

Exc Voltage / 2<43.3 to 5.5<100>5030100%
Isolated from Mounting Surface

ENVIRONMENTAL
Temperature Response (%)
Operating Temperature (°C)
Storage Temperature (°C)-10/+20 from -20°C to +80°C-20 to +80-20 to +80

PHYSICAL
Sensing Element
Case Material
Weight (grams)

Ceramic (shear mode)
Ceramic Base, Nickel Silver Cover
3.6Notes±5025.02-2000>10000±2<105000
21012080
40±10012.52-2000>10000±2<10500016012080
40±2006.252-2000>10000±2<10500015012080
40±5002.52-2000>10000±2<105000160
400320160±30%±2dB

2Hz-10kHz@ 10Hz@ 100Hz@ 1000Hz@100Vdc

1A wider frequency response of 2-6000Hz is available on model 832M1
The model 832 is not to be reflow soldered at high temperature, manual soldering is recommended. See operating manual.
3
The model 832 can be operated with 2.8V excitation but the full-scale range will be limited. See operating manual for details.
2Calibration supplied:

CS-SENS-0100

NIST Traceable Amplitude Calibration at 80Hz

The information in this sheet has been carefully reviewed and is believed to be accurate; however, no responsibility is assumed for inaccuracies. Furthermore, thisinformation does not convey to the purchaser of such devices any license under the patent rights to the manufacturer. TE Connectivity reserves the right to makechanges without further notice to any product herein. TE Connectivity makes no warranty, representation or guarantee regarding the suitability of its product for any
particular purpose, nor does TE Connectivity assume any liability arising out of the application or use of any product or circuit and specifically disclaims any and all
liability, including without limitation consequential or incidental damages. Typical parameters can and do vary in different applications. All operating parameters must
be validated for each customer application by customer’s technical experts. TE Connectivity does not convey any license under its patent rights nor the rights ofothers.

SENSOR SOLUTIONS /// Model 832 Rev H

12/2017

Page 2

MODEL 832 ACCELEROMETERDIMENSIONSSCHEMATICSENSOR SOLUTIONS /// Model 832 Rev H

12/2017

Page 3

MODEL 832 ACCELEROMETER

ORDERING INFORMATION
832

GGGG

Range
0025=25g
0050=50g
0100=100g
0200=200g
0500=500g

Example;832-0500
Model 832, 500g range

TE.com/sensorsolutions
MEAS France SAS and Measurement Specialties (China) , Inc., are TE Connectivity companies.
TE Connectivity, TE, TE connectivity (logo) are trademarks. All other logos, products and/or company names referred to herein might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes nowarranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard Termsand Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuse of theproduct. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2017TE Connectivity Ltd. All Rights Reserved.

SENSOR SOLUTIONS /// Model 832 Rev H

12/2017

Page 4