Uf 801-P

FULL
PIPE

LIQUID & GAS
APPLICATIONU LT RAS ONIC PORTABLE FLOW M ETER

MEDIA
MEASUREDLIQUIDS& GASES

MODELSTANDARD
DUAL PIPEDUAL CHORDPIPE DIAMETERS
UP TO
10 000MM

COMPACT›L
 ight weight(less than 1kg)›E asy to use

ROBUST› IP68 ABS enclosure

CALORIMETERDUAL CALORIMETERADVANCEDFUNCTIONSHIGHPERFORMANCE

MULTIPLE USES›M
 ulti-parameter data logger›G raphic screen›S tores up to 11 configurations/sites›T imer/programmer›E cho, gain and quality
index displayed›O ptional Input/output modules(analogue, digital)›B attery life up to two months, using
timed operation› On every type of homogeneous
liquid, even non-conductive› On most types of gases - high andmedium pressure*› Non ideal flow conditions taken
into account

RELIABLE› Automatic zero calibration› Ten flow calculations
per second

TYPICAL APPLICATIONSDrinking water:
Leakage detection,
pump flow control,
control of in-line flow meters
Water (raw, waste):
Pump flow control
Flow surveys: Troubleshooting
installations, resolving disputes
Civil engineering:
Validation of system performance before
handover of a project
Climate engineering: System balancing,thermal assessment
Hydrocarbons:Temporary flow measurement* APPLICATION CONDITIONS: PLEASE CONTACT US

Ultraflux
U l t ra s o nic flowmetersUf 801-P
MODEL
REFERENCE

STANDARD

DUAL PIPEDUAL CHORDCALORIMETERDUAL CALORIMETER

CO_801POR1BINAB___

CO_801POR2BINAB___

CO_801POR2BINAB___

CO_801CAL1BINAB___

CO_801CAL2BINAB___TECHNOLOGY

Portable ultrasonic transit-time flowmeter - Continuous and bidirectional flow metering - 10 flow measurement/s

SIGNAL ANALYSIS

By Digital Signal Process (real-time Echo Shape Control, digital filtering and regulation of gain on each firing)

ACCURACYREPEATABILITY
LINEARITY

Up to 0.5 % of reading - minimum velocity 0,2m/s for pipes above 300mm
Up to 0,1%
Up to 0,1 %VELOCITY RANGE+/- 30 m/s

TEMPORAL RESOLUTION

0,1 nsRESPONSE TIME

Less than 1 second

DAMPING

Adjustable from 0 to 3600 s

INTERNAL Ø OF PIPE

From 8mm to 9,900mm approximately (depending on pipe thickness)

EXTERNAL Ø OF PIPEFrom 10mm to 10,000mm

PIPE MATERIAL

Aluminium, asbestos, cast iron, copper, glass, grey cast iron, nylon, Plexiglas, polyethylene, PTFE, PVC, stainless-steel and steel. Other materials can be used if their physicalproperties are known.

MULTI LAYER PIPE MATERIAL

Aluminium, asbestos, cast iron, copper, glass, grey cast iron, nylon, Plexiglas, polyethylene, PTFE, PVC, stainless-steel and steel. Other materials can be used if their physicalproperties are known.

STANDARD INPUTS/OUTPUTS—TEMPERATURE INPUTSFOR ENERGY CALCULATION- DUAL MODULE -—TEMPERATURE INPUTSFOR ENERGY CALCULATION(DUAL CALORIMETRY)- DUAL MODULE --————

PT100/PT1000 2-input module
taking up the physical space oftwo modules

USE

Flow measurement

Flow measurement in two pipes

Flow measurement with two
speed chords

Flow measurement
and calorimetry

Flow measurements in two pipes
and dual calorimetry

SINGLE OR DUAL PIPE

Single pipe

Dual pipe

Single pipe

Single pipe

Dual pipe

SINGLE OR DUAL CHORD

Single chord

Single chord

Dual chord

Single chord

Single chord

PIPE THICKNESS MEASUREMENT

Function available in digital and graphical mode(SE-1701)

PT100/PT1000 2-input module
taking up the physical space of twomodules

Up to 4 modules to choose from:IN OP T I ON ,S UP P L E M E N TA RY I NP U T/OU T P U T S I N G LE MODULES

Up to 2 modules to choose from:› 1 isolated, active analogue output: current 4-20mA, 0-20mA, 0-24mA, maximum load impedance 750 Ω • Module 1› 2 static relay outputs (50V - 10mA) usable as frequency outputs (up to 1kHz) • Module 2›2
 isolated current inputs 4-20mA, 0-20mA, 0-24mA • Module 3›2
 0-10V voltage inputs • Module 4›2
 contact inputs (pulse or state) • Module 6—DISPLAY› Graphical LCD screen (14 lines x 20 characters)› Backlit screen with time delay feature› Flowrate unit: l/s, l/min, l/h, m3/s, m3/h, m3/day, Gps, Gpm, Gph, Bps, Bpm, Bpd

MEAS U R E D VA LU ES

Volumetric flowrate, fluid velocity and speed of sound - Totalizers: 4 independent and adjustable
Signal quality analysis: strength, quality index and shape (via the oscilloscope function)

TROUBLESHOOTING HELP

Oscilloscope function (echo displayed) • Gain • Quality index

SET-UP› Quick and simple - uses 7-key touch pad with 2 for dynamic allocation - or via dedicated software supplied› Possible to build in an access code

MEASUREMENT DAMPING TIME

From 0 to 3600 seconds

INFORMATION STORAGE› 4MB data logger: time stamping - between 1 and 30 variables - up to 266,706 lines› 3-variable time stamping: 133,353 lines • 14 variables: 35,560 lines • 30 variables: 17,206 lines› Logging frequency from 1 second to 24 hours

CONFIGURATION RECORDING

Up to 11 embedded configurations

TOTALIZATION

Resolution from 1 ml to 1000 m3

OPERATING SYSTEM

Ultraflux dedicated software (Windows compatible) for configuration (upload/download the settings), read/record the measurement values and download the logger’s data.
Measured values and logged data are readable with spread sheet software (Microsoft Excel, etc.)

PROGRAMMERProgrammable power-up to increase the logger’s battery life

2/3 LANGUAGES

English & Russian or French & English + 1 additional language to be chosen: German • Portuguese • Spanish • Italian

BATTERY LIFE

Up to 14hr continuous use • Charge indicator

COMM U N I CAT I ON

Serial link RS232 or RS485 to JBUS/MODBUS protocol • 115,200 Bauds - USB port

ACCESSORY INCLUDED

1 RS232 to USB converter link cable

ELECTRICAL CHARACTERISTICS› 12V NiMh sealed battery› Charger with input: 100-240V ac / 1.05-0.55A / 47-63Hz and output: 18V / 2.5A”› Cable for auxiliary power supply available as an option

ENCLOSUREABS • 900g • 220 x 115 x 64mm

PROTECTIONEN/IEC 60529 IP68

COMPLIANCE

EMC compliance: EN/IEC 61326-1 - Safety compliance: EN/IEC 61010-1

TEMPERATURE RANGEFor use from -10°C to 50°C

PROBE CABLE

Ultraflux cable - Standard: 5 meters - Optionnal: up to 100 meters

NON CONTRACTUAL DOCUMENT

Ultraflux