CE marking (EMC EN61326-1:2013, RoHS EN50581:2012)*1

The world's fastest response time of 0.1ms(0.0001s)
by using Indium antimonide(InSb).
*1 small type also available:TMHX-CHE (1ms)*2 at room temperature (20 〜 25℃)

New standard model with additional functions.

Usable at the long distance of 500mm.
Wide temperature range : 0〜1350℃*3

Usable for a very small spot size of φ0.7mm.

*3 small type available:measurement distance 100mm & spot size Φ1mm.
●Aluminium body with robustness, heat resistance and chemical resistance is equivalent to IP67.
●We have more various thermometers for low-temperature metals, mirror surface, glass, film, through quartz, lamp heating and flame detection.
Please check the specifications sheet.

2/6Performance remarkably improved withindium antimonide(InSb).
Quick response time with direct conversion of infrared rays.

Generally, a thermopile element converts infrared rays into thermal energy first, and then into electricalenergy. Therefore, its response time is constrained. However, our the TMHX series using indium antimonide(InSb) is able to convert infrared rays directly into electrical energy and realizes the world's fastest responsetime for measuring at room temperature (20 〜 25℃).
Thermopile method(conventional products)

Infrared rays

TMHX series

Infrared rays

Heat

Thermocouple

Voltage

Direct conversion

Current

TMHX-CHE=1ms（0.001s） TMHX-CNE/CLE/CSE=10ms（0.01s）
Response time

Temperature (℃)

600

1mS

500
400
300
200
100
0
0

1

2

3

4

5

Time (ms)Stability remarkably improved.
The TMHX series using indium antimonide(InSb) remarkably improves the stability of measurementcompared to the previous model, the TMH91, which uses a thermopile element(check the chart below).
The TMHX series uses a short wavelength 2 〜 6.8µm, which gets almost twice the emissivity measurement
values for metals(iron, SUS, chrome, etc.) and realizes more stability in the measurement.

Stability (Reference values)

Temperature (℃)

102

TMHX-CHE

101

TMHX-CNE

100

TMH91

99

98

0

50

100

Averaging time 0.05s. Emissivity 1.0.

150Time (ms)

3/6

200

Pursuing usability and durability.
Easy alignment with "the optical coaxial LED sight"
It is critical to accurately focus on a target in order to get a precise measurement.
The TMHX series uses "the optical coaxial LED sight" that can emit the visible light to sight on a target foreasy alignment.
Sensor head

The optical coaxial LED sight
Measurement area

Laser single sight
LED sight light

Sensor head

The single sighting cannot measure the wholemeasurement area since it can indicate only the centerof the target area.
Sensor head

Dual sights

LED sight light

LED sight light

The dual sighting cannot measure the wholemeasurement area since it can indicate only bothsides of the target area and the focus error
becomes large once the focus is lost.

This can indicate most of the target area, so that it iseasy to adjust almost the whole measurement area
as the same way as human eyes.

Measuring a very minute point of ∅0.7mm.
The TMHX-CS can measure a very minute spot size down to a diameter of 0.7mm. You can choose thetarget size from 0.7mm to 9.0mm for your application.
Images of the actual dimensions
TMHX-CSE ∅0.7mm

TMHX-CNE∅1.6mm

TMHX-CHE∅3.5mm

TMHX-CLE ∅9.0mm

Aluminium body ensures a high durability equivalent to IP67.
The TMHX series uses an aluminium body that improved robustness, heat resistance and chemical
resistance (compliant with the IP67 rating), equivalent to IP67. The size is approximately 30% smaller thanthe previous model the TMH91. The symmetric design is easy to install in various locations and the new
cable has also improved durability.
The body is approximately
30% smaller than the previous
model the TMH91.

※IP67

Ingress Protection ratings defined in international standard EN 60529.

30mm

The first number - Protection against solid objects.
48.5mm6: Protected against dust that may harm equipment.
The second number - Protection against liquids.

7: Protected against the effects of temporary immersion between 15cm
and 1m. Duration of test 30 minutes.

4/6

Supporting various applications.
The TMHX series has various displays and parameter setting units to meet your requirements such as the way toshow the temperature, input emissivity and data communication. You can configure the systems by yourself with
the TMHX series and the equipment you have. Please check the display guide and the display specifications
sheet.

Display : TMCX-NDE

Analog output
Alarm output
Emissivity input

RS232C
Analog

Display : TMCX-HA/HD

Analog output
Alarm output
Emissivity input
RS485

RS232C
AnalogAnalog output
Alarm output
microSD card

Touch screen type : TMCX-TDE
RS232C
AnalogParameter setting unit : PWCXRS232C

Batteries type

Software for parameter setting : PWSX
RS232C

USB-RS232C converter : PWUXRS232C

Temperature Adjustment

Data logger

PC

Recorderprogrammable logic controller

5/6Specifications
TMHX-CNE0500-0070E003

TMHX-CNE0500-0035E1.6

Model

Temperature
range

TMHX
-CNE0500-0120E5.5

TMHX-CUE0500-0200H008

TMHX-CLE1350-0500B009

TMHX-CSE0500-0040H0.7

0〜500℃

0〜1350℃

0〜500℃

3〜5.6μm

5〜6.8μm

1000℃
500℃
0℃

0〜500℃

Spectral range
Measurement
distance

2〜6.8μm
35mm

70mm

120mm

200mm

500mm

40mm

∅ 1.6mm∅ 3.0mm∅ 5.5mm∅ 8.0mm∅ 9.0mm∅ 0.7mmSpot size
10ms（ 0.01s）〜

Response time

0.1ms（ 0.0001s）〜

Analog output

4〜20mA／0〜20mA／0〜1V／mV/℃ (switchable)

Alarm output

open-drain: 27VDC, 0.2A

Communications

RS232C (non-isolated)

Supply voltage

DC4.7〜27V、0.1A max.

10ms（ 0.01s）〜

CE mark certified (EMC EN61326-1:2013, RoHS EN50581:2012)

Accessories

Mounting
Shield casebracketProtective
Shield casewindow

TMSX-ATMSX-B4

ShieldShieldcase
case

TMAX-ATMSX-ATMSX-B4
TMAX-B

TMDX-A1CTMSX-ATMSX-B4
TMDX-15C

Use
with thesensorInfluence
reduction
head
protect fieldfromof the tomagnetic
for high frequencyat
electromagnetic
heating
applications.

BracketsensorInfluencefor
reduction
headof the installation.
magnetic fieldUsewith reduction
the lens toUsewith thelens toInfluence
Influence
reduction
prevent
dust andfield
prevent
contaminationof the magnetic
fieldof the magnetic
condensation.
for high
frequency
for high frequency
anddamage.

Water
Shieldcooling
casejacket

Extension
Shield case
cable

TMWX-A1TMSX-ATMSX-B4
TMWX-B4

TMBX-E05
TMSX-ATMSX-B4

for high frequency
heatingUse
for installing
sensor Extension
cable:5m
Influence
reduction
Influence reduction
heads
where thefielde-CONof the magneticof the magnetic field
for high frequency
for high frequency
ambient
temperature isheatingheatingmorethan 50℃

ShieldAir purge
casefood
TMPX-A1TMSX-ATMSX-B4
TMPX-25

heatingheating

ShieldBranchcase
cable

ShieldResistorcase

Airless
Shielddust
case
protector Right
Shieldangle
casemirror
TMNX-A1TMSX-ATMSX-B4
TMNX-25

TMLX-A1CTMSX-ATMSX-B4
TMLX-15C

Use
with thelens toInfluence
reduction
prevent
dust. fieldof the magneticNoairsupplyrequired.
for high frequency

Reflects
sensorInfluence reduction
fieldview byof the ofmagnetic
field90
for high frequency
degrees.

ShieldCapacitor
case

ShieldFerritecase
coreheatingheating

TMBX-B01
TMSX-ATMSX-B4

TR-251NTMSX-ATMSX-B4
TR-501N

TC-105N
TMSX-ATMSX-B4

FC-2032
TMSX-ATMSX-B4

For
setting
unit
Influence
reduction
whenthermometerof the magnetic
field
forusedhigh frequency
isalone.
heatinge-CON branch.

For
the analog
outputInfluence
reduction
0-5V/10Vof the magnetic fieldFor
analogoutputInfluence
reduction
noise.
Connect
toof the magnetic
field
for high frequency
analogsignalheating
receiver.

For
power
noise.
Influence
reduction
Installto cables.
of the magnetic
field

for high frequency
heating

for high frequency
heating●The contents described in this catalog are subject to change for improvement without notice.

●If you have any questions, contact us at the address orlinks shown below.

http://www.japansensor.co.jp
info@japansensor.co.jp

6/6

TMHX-Cata-V201906P201906 1K

Features

General purpose

Minute points

Long distance:500mm

Photo

Model

TMHX
-CNE0500-0035E1.6

TMHX-CNE0500-0070E003

TMHX-CNE0500-0120E5.5

TMHX-CSE0500-0100E001

TMHX-CSE0500-0040H0.7

TMHX-CLE1350-0500B009

1,000℃

Temperature
range

500℃
0℃

2.0 〜 6.8μm

Spectral range
Measurement distance0 〜 500℃

0 〜 500℃

0 〜 1350℃

2.0 〜 6.8μm

5.0 〜 6.8μm

3.0 〜 5.6μm

35mm

70mm

120mm

100mm

40mm

500mm

∅ 1.6mm∅ 3mm∅ 5.5mm∅ 1mm∅ 0.7mm∅ 9mm

Spot size*2

Accuracy*1

Repeatability*1

Resolution*1< 300℃:±3.0℃≧ 300℃: measured value ±1%

± 5℃

±0.5℃

≦ 0.5℃

0.01 〜 5s
Response Time （0 〜 95％ analog output) configurable bysmoothing function< 300℃:±3.0℃<300℃:±3.0℃≧ 300℃: measured value ±1% ≧ 300℃: measured value ±1%

±2℃

±1℃

±1℃

< 100℃: ≦ 3℃
≧ 100℃: ≦ 1.5℃

< 50℃: ≦ 1℃
≧ 50℃: ≦ 0.5℃

< 50℃: ≦ 1℃
≧ 50℃: ≦ 0.5℃

0.05 〜 5s（0 〜 95％ analog output)
configurable by smoothing
function

0.01 〜 5s（0 〜 95％ analog output)
configurable by smoothing
function

0.01 〜 5s（0 〜 95％ analog
output) configurable bysmoothing function *3

Warm-up time

1min

1min

3min

1min

Dimension category

HX-C1

HX-C2a

HX-D4

HX-C2bWeight(cable not included)

80g

85g

190g

90g

2m Direct leading out

2m Direct leading out

2m With a connector

2m Direct leading out

A

B

C

B *4

Cable
Accessory
category*1

Ambient temperature 23±5℃, Emissivity 1.0, Averaging time 0.05s.

*2

Check the optics diagrams for spot sizes outside of the ranges in the measurement distance section above. It shows the area
where 90% of the total energy is included.

*3

The response time may be longer by 0.001s when switching the internal range.

*4

Water cooling jacket the TMWX-A3 is available.

1/12

Infrared Thermometer TMHX-C series Specifications
Features

Quick response : 1ms

Ultra quick response: 0.1ms

Glass

Photo

Model

TMHX
-CHE0500-0100B3.5

TMHX-CHE0500-0200B007

TMHX-CUE0500-0200H008

TMHX-CGE1200-0180E007

TMHX-CGE1200-0500E020

TMHX-CGE2400-0150H2.2

TMHX-CGE2400-0300H4.5

TMHX-CGE2400-1500H020

2000℃

Temperature
range

Spectral range
Measurement distance

1000℃
0℃

0 〜 500℃

0 〜 500℃

2.0 〜 6.8μm

2.0 〜 6.8μm

50 〜 1200℃

50 〜 2400℃
5.0 〜 5.6μm

100mm

200mm

200mm

180mm

500mm

150mm

300mm

1500mm

∅ 3.5mm

∅ 7mm

∅ 8mm

∅ 7mm

∅ 20mm

∅ 2.2mm∅ 4.5mm∅ 20mmSpot size*2

Accuracy*1

Repeatability*1

Resolution*1

Response Time

< 300℃:±3.0℃≦ 300~1200℃: measured value ±1%≧ 1200℃: measured value ±2% *5< 400℃:±4.0℃< 350℃:±3.5℃< 300℃:±3.0℃≧ 400℃:≧ 350℃: measured value ±1%≧ 300℃: measured value ±1%
measured value ±1%
±1℃

±0.5℃

Measured value ±0.2% ±2℃±1℃< 50℃: ≦ 1.5℃
≧ 50℃: ≦ 0.7℃

< 50℃: ≦ 4℃
≧ 50℃: ≦ 3℃

< 100℃ : ≦ 1℃
≧ 100℃ : ≦ 0.5℃

0.001 〜 5s
(0~95% analog output)
configurable by smoothing
function

0.05 〜 5s
0.0001 〜 5s
(0~95%
analog output)(0~95% analog output)
configurable by smoothing
configurable bysmoothing function function*30.01 〜 5s(0~95% analog output) configurable bysmoothing function*4Warm-up time

1 min

3 min

1 min

3 min

Dimension category

HX-C2a

HX-D4

HX-C1

HX-D4

Weight(cable not included)

85g

190g

80g

190g

2m Direct leading out

2m With a connector

2m Direct leading out

2m With a connector

B

C

A

C

Cable
Accessory
category*1

Ambient
Ambient temperature 23±5℃, temperature
Emissivity 1.0,
23±5℃, Emissivity
Averaging time 0.001s.
1.0, Averaging time
0.0001s.

*2

Check the optics diagrams for spot sizes outside of the ranges in the measurement distance section above. It shows the areathat 90% of the total energy is included.

*3

The response time may be longer by 0.001s when switching the internal range.

*4

The response time may be longer by 0.002s when switching the internal range.

*5

Accuracy not guaranteed for the temperature higher than 2,000℃.

Ambient temperature 23±5℃, Emissivity 1.0, Averaging time 0.05s.

2/12

Infrared Thermometer TMHX-C series Specifications
Features

Mirror surface

Film

Through quartz

Lamp heating

Through
flame

Flame
detection

TMHX-CPE1200-0180E007

TMHX-CVE1300-0200H4.5

TMHX-CBE1500-0300E018

0 〜 1200℃

200 〜1300℃

150 〜1500℃

5.0 〜 6.8μm

3.9μm

4.3μm

180mm

200mm

300mm

7mm

∅ 4.5mm∅ 18mm

Photo

Model

TMHX
-CME0250-0020E004

TMHX-CME0250-0050H003

TMHX-CFE0350-0200E012

TMHX-CQE0500-0200E011

TMHX-CQE0500-0200H004

1500℃

Temperature
range

1000℃
500℃
0℃

50 〜 250℃

80 〜 350℃

2 〜 6.8μm

3.4μm

Spectral range
Measurement distance

20mm *Note550mm *Note5

200mm

∅ 4mm

∅ 3mm

∅ 12mm50 〜 500℃
3.0 〜 4.0μm

2.8 〜 3.9μm

200mm

Spot size*2

Accuracy*1±5℃

Repeatability

±2℃

*1

Resolution

Dimensioncategory
Weight(cable not included)
Cable
Accessory
category

±4℃

±1℃

±1℃

< 100℃: ≦ 1℃
≧120℃: ≦0.5℃ ≧100℃: ≦0.5℃

0.05 〜 5s(0~95% analog output)
configurable by smoothing
function

3 min

Warm-up time

∅ 4mm

< 300℃:±4.0℃< 300℃:±3.0℃≧ 300℃:≧ 300℃:
measured value
measured value ±1%±1.5%< 120℃: ≦ 1℃

≦ 2℃

*1

Response time

±4℃

∅ 11mm< 300℃:±3.0℃< 300℃:±3.0℃ ≧ 300~1200℃:
measured value ±1%≧ 300℃:
measured value ±1% ≧ 1200℃:
measured value ±2%

none

Measured value±0.2% ±0.2℃

±5℃

±0.5℃

≦ 0.5℃

0.05 〜 5s
0.05 〜 5s(0~95% analog(0~95% analog output)
output) configurable configurable by smoothingby smoothing
function*3function≦ 0.5℃

0.01 〜 5s(0~95%
analog output)
configurable
by smoothing
function*3< 300℃: ≦ 1℃

≧300℃: ≦0.5℃≦ 1℃

0.01 〜 5s
0.05 〜 5s(0~95%
analog output) (0~95%
configurable analog output)
by smoothing configurable bysmoothing function
function
*4

1 min

1 min

3 min

1 min

3 min

1 min

HX-C1
＋ Light shielding hood

HX-D5

HX-C1

HX-C1

HX-D4

HX-C1

HX-D4

HX-C1

110g

210g

80g

80g

190g

80g

190g

80g

2m Direct leading out 2m With a connector 2m Direct leading out 2m Direct leading out 2m With a connector 2m Direct leading out 2m With a connector 2m Direct leading out
A

C

A

A

C

A

C

A

*1

Ambient temperature
Ambient temperature 23±5℃,
Ambient temperature 23±5℃,
23±5℃,
Emissivity 0.05,
Emissivity 1.0,
Emissivity 1.0,
Averaging time 0.5s.
Averaging time 0.1s.
Averaging time 0.5s.

*2

Check the optics diagrams for spot sizes outside of the ranges in the measurement distance section above. It shows the areathat 90% of the total energy is included*3

The response time may be longer by 0.001s when switching the internal range.

*4

The response time may be longer by 0.002s when switching the internal range.

*5

Valid measurement distance; ±10mm.

3/12

Ambient temperature Ambient temperature Ambient temperature
23±5℃,
23±5℃,
23±5℃,
Emissivity 1.0,
Emissivity 1.0,
Emissivity 1.0,
Averaging time 0.05s. Averaging time 0.05s. Averaging time 1s.

Infrared Thermometer TMHX-C series Specifications

Common Specifications
Detective Element

InSb (Indium antimonide)

Sight

Red LED sight *switchable on/off
Analog output (non-isolated)
Output type (selective)

Output Range

Output Accuracy *1

0 〜 1V≧ 30mV±1.5mVmV/℃≧ 30mV（30℃）±1.5mV0 〜 20mA≧ 0.2mA±0.02mA

4 〜 20mA≧ 4.0mA±0.02mA

Output

RS232C output (non-isolated)
output swing: approx.±4V

baud rate: 4800, 9600, 19200, 38400, 57600, 115200bps
Alarm output (non-isolated)
open-drain: 27VDC, 0.2A
hysteresis: 0 ~ 99.9℃
Peak Hold

Emissivity correction

reset(switchable)

time: 0.01 ~ 10s (configurable)

discharge: time 0.01 ~ 10s, level 0.2 ~ 1.0

guaranteed range ; 0.3 〜 1.0 *2
setting range ; 0.05 〜 1.000 (setting resolution:0.001)◎ reflected correction function

Sensor correction function

span:0.500 ~ 2.000 / zero: -50 ~ +50( ℃ or ℉ selectable)

Temperature indication

none

Cable length

2m (standard)Protection classsame quality as IP67, not included:e-con connector
0 〜 50℃

Ambient temperature
Ambient humidity*330 ~ 85%RH (non-condensing)−15 〜 70℃

Storage temperature*3

DC4.7 〜 27V、0.1A max.

Supply voltage
Aquired CE marking (EMC EN61326-1:2013,RoHS EN50581:2012)*1 : Static error is reflected in the accuracy rating.

*2 : The TMHX-CME, guaranteed range is 0.05 〜 1.0.

*3 : The TMHX-CME250-0020E004, ambient temperature 0 〜 70℃, storage temperature -15 〜 80℃.

Ordering information

Cable length-00

2m (standard)

-05

5m

-10

10m

Analog output
mV/℃
4 - 20mA
0 - 20mA
switchable

4/12

0

0 〜 1V (standard)5

0 〜 5V (built-in resistor)

1

0 〜 10V (built-in resistor)

Infrared Thermometer TMHX-C series Specifications

Operations
Alarm outputs
Mode No.

Measurment temperature

Alarm name1

High ON

2High OFF3

Low ON4

Low OFF

5

Band ON

6

Band OFF7

Error ON

8

Error OFF⇦ Low

Low set point

High set point

High ⇨ON*Error : an inner voltage malfunction

Peak-hold operation
Hold time

Hold time

Hold time

Input value

Hold output
Peak-hold (Reset time)

Input
Time

Discharge time
1.0
Input value

Hold output
Discharge
level

Peak-hold (Discharge reset)

Input
0

Time

Input value

Averaging time

Averaging
Input signal

Time

5/12

Infrared Thermometer TMHX-C series Specifications

Optics diagram
TMHX-CNE0500-0035E1.6

It shows the area that 90% of the total energy is included.

Ø5

Ø14

Ø9

Ø5
Ø1.6

TMHX-CNE0500-0070E003

Ø5

TMHX-CNE0500-0120E5.5

Ø5

Ø7

Ø4

Ø18

Ø13

Ø3

Ø13

Ø9

Ø5.5
Ø5.5

Ø7.6

TMHX-CSE0500-00100E001

Ø4.3

Ø5.2

Ø17.8

Ø13.6

Ø9.4

Ø1
Ø23

TMHX-CSE0500-0040H0.7

Ø12

Ø13

Ø35

Ø24

Ø0.7

Ø13

TMHX-CHE0500-0100B3.5

Ø19

Ø12

Ø8
Ø3.5

Ø13

Ø12

TMHX-CHE0500-0200B007

Ø10

Ø17

Ø12

Ø9
Ø7

Ø22

Ø23

Ø15

TMHX-CUE0500-0200H008

Ø8

0

35
40 50 60 70 80

20

100

120

150

200

250

300

Measurement distance (mm)Ø13

TMHX-CLE1350-0500B009

Ø12

Ø14

Ø10

Ø23

Ø31

Ø40

Ø49

Ø58

Ø66

Ø75

Ø9

0

200

400 500 600

800

1000

1200

Measurement distance (mm)

6/12

1400

1600

1800

2000

Infrared Thermometer TMHX-C series Specifications

Optics diagram

TMHX-CGE1200-0180E007

TMHX-CGE1200-0500E020

It shows the area that 90% of the total energy is included.

Ø7.5 Ø7.3

Ø33

Ø25

Ø17

Ø9

Ø7

Ø47

Ø37

Ø7

Ø22

TMHX-CGE2400-0150H2.2

Ø16

Ø11

Ø9

Ø26

Ø18

Ø20

Ø2.2
Ø23

TMHX-CGE2400-0300H4.5

Ø17

Ø11

Ø23

Ø14
Ø4.5

0

50

150
100 180 200 250 300

Ø23

TMHX-CGE2400-1500H020

400

500

600

800

Ø49

Ø35

Ø21

Ø22

1000

Ø20

0

1000

1500

Ø12
Ø23

TMHX-CQE0500-0200H004

Ø14

2500

Ø43

Ø30

Ø17

2000

Ø42

Ø32

Ø22

Ø11

Ø9

TMHX-CFE0350-0200E012

Ø4
Ø39

Ø30

Ø21

Ø10

Ø9

TMHX-CQE0500-0200E011

TMHX-CPE1200-0180E007

500

Ø11
Ø7.5

Ø7.3

Ø33

Ø25

Ø17

Ø9
Ø7

Ø22

TMHX-CVE1300-0200H4.5

Ø14

Ø43

Ø31

Ø18
Ø4.5

Ø36

Ø9

TMHX-CBE1500-0300E018

Ø79

Ø61

Ø18

0

50

100

180 200 250 300

400

500

600

800

Measurement distance (mm)

Ø20
Ø8

Ø7

TMHX-CME0250-0050H003
0

10

20

30

40

Ø5.6 Ø4.8 Ø6.4

TMHX-CME0250-0020E004

Ø3

1000

Ø4

50

60

70

Measurement distance (mm)

80

90

7/12

100

0

10

20

30

40

50

Measurement distance (mm)Infrared Thermometer TMHX-C series Specifications

Sensor head dimensions（mm）
HX-C1
24

17

Cable length (standard 2m)

Ø7

4

12

6

18

15

24.5

20

1
8

Connection cable
Mounting hole2-Ø3.5 Counter bore Ø6 depth 3

Lens

12

30

48.5

17.6

Measurement
distance

16

Weight: Sensor head 80g, Cable 150g

HX-C2b

30

17.6

15

Ø20

36

12 12

Mounting hole2-Ø3.5 Counter bore Ø6 depth 3

84.5

48.5

4

24

17

12 12

30

24

17

20

4

Measurement
distance

15

48.5

20

24

72.5

Ø20

Measurement
distance

17.6

HX-C2aMounting hole2-Ø3.5 Counter bore Ø6 depth 3

16Weight: Sensor head 85g

Weight: Sensor head 90g

HX-C2a / HX-C2b Alternative cable

Cable length (standard 2m)

18

Ø7

17

6
24.5

1

8

Weight: Cable 150g

HX-D4
Measurement distance

104

44

53

51
42

42

6

22.7

23

33

Ø29.8

46

Ø15

5

30

15

Lens

Lens barrel

Mounting hole3-Ø3.5 Counter bore Ø6 depth 3

Cable length (standard 2m)

18

1

24.5

Ø7

15

8
Head connector

Connection cable

Weight: Sensor head 190g, Cable 160g

8/12

16

Infrared Thermometer TMHX-C series Specifications

Sensor head dimensions（mm）
HX-C1+Shading hood
24
18

1

20

Ø18
15

12

6

8

Connection cable
Mounting hole2-Ø3.5 Counter bore Ø6 depth 3

Light shielding hood
(TMFX-10-25 option)
Lens

12

16

Weight: Sensor head 80g, Light shielding hood (TMFX-10-25) 30g and Cable 150g

HX-D5
Measurement distance

Cable length(standard 2m) 18

Headconnector

24.5

Connection cable
15

Mounting hole3-Ø3.5 Counter bore Ø6 depth 3

1

46

22.7

33
6

Lens
barrel
Shading board
TMVX-25-50
Lens

Ø7

Ø15

5

23

Ø86

44

51
42

42

1

104

Ø29.8

2

53

8

Head connector

15
30

6

Weight: Sensor head 210g, Cable 160g

Wiring (in case the thermometer is used alone)
Sensor head TMHX

Connection cable
DC+IN

outsidesighting
input

3.3V

TXD
GND

Alarm
outputRXD

Output connector
e-CON 8P

Red
DC+4.7
to 27V

Yellow
Black

Analog output
0 〜 1V0 〜 5V0 〜 10VmV/℃
4 〜 20mA
0 〜 20mA

RXD
RS232CSG

GND

OrangeTXD

White

AnalogoutputIn case of outsidesighting

GreenR

Purple
A.GND

Blue

9/12

Minimum power supply voltage
more than 4.7V
more than 8V
more than 13V
more than 4.7V
As per terminator
resistance

Load
Analog output

DC+2
to 27VIn case of alarm output

Attach in case of 0〜5V or 0〜10V
Resistor for 0〜5V (TR-251N)
Resistor for 0〜10V (TR-501N)
Unnecessary in case of built-in resistor

30

Cable length (standard 2m)

4

17.6

17

Ø7

25

48.5

24.5

Measurement distanceInfrared Thermometer TMHX-C series Specifications

Accessories
TMAX-A

Mounting bracket

TMAX-B

Accessory : A,B

Bracket for sensor head installation

Accessory : C

Mounting bracket
Bracket for sensor head installation

t 2.0
Weight : 45g

16.6

2
2C

Ø4Hole

4C

42
33

39

6.5

59

°

3-10×4
Long hole

90

20
50

24

54

3

4

6

20

20

2

6

24

12

C
4-

5

4

4 8 14
Ø4 Hole27

Hole

2-10×4Long hole 2-Ø4Hole

3 8

Hole

7

4

Ø4 Hole

37

t 2.0

13

°

90

16

Hole

78

32

Protective window TMDX-A1C / TMDX-15C / TMDX-25C

TMPX-A1 / TMPX-15 / TMPX-25

Air purge hood

Use with the lens to prevent contamination and damage.

Weight : 85g

32

Use with the lens to prevent dust and condensation.
Ø4 tube fitting

Instrumentation
air entrance

B

C Nut
D

C
Model

TMDX-A1C
TMDX-15C

TMDX-25C

A
Ø20.5

B
7

Ø20.5

7

Ø30.5

9

C
12
10

12

D
M16P2

Accesories

M16P0.75

M26P0.75

E

B

A

A

D

F

Window materials

Model
A
TMPX-A1 Ø18
TMPX-15 Ø20

A
B

TMPX-25

C

Airless dust protector TMNX-A1 / TMNX-15 / TMNX-25

Ø30

B
Ø10

C
35

D
40

E
M16P2

F
24.5

Accesories

Ø25

75

79

M26P0.75

32

C

Ø17

Right angle mirror

Use with the lens to prevent dust. No air supply required.

50

53

M16P0.75

AB

27

TMLX-A1C / TMLX-15C / MLX-25C

Reflects sensor field of view by 90 degrees.
A

Fixed screw

Reflection mirror

D

E
G

D

Diaphragm

A

O ringTurn

B
C

TMNX-A1

Model

A
Ø28

B
50

C
54.5

D
M16P2

Accesories

TMNX-25

Ø66

107

112.5

M26P0.75

C

TMNX-15

Ø40

70.5

73.5

M16P0.75

A
B

Model
A
B C D
E
TMLX-A1C 23.8 20 5 14.9 M16P2
TMLX-15C 23.8 20 3 14.9 M16P0.75

TMLX-25C

10/12

36

30

5 22

M26P0.75

F

B

C

O ring

Window materials

F G Optical path Accesories
34 29.8
34.9
A
34 29.8

50 45

34.9

52

B

C

Infrared Thermometer TMHX-C series Specifications

Accessories
TMSX-A
Accessory : A,B

36

58 ±0.5

TMWX-A1 / TMWX-A2 / TMWX-A3

Water cooling jacket

Use with the sensor head to be installed where theambient temperature is above 50℃.

Air controller
for the container purge air inlet
dried air for 4×2.5 tube
(for avoiding condensation)

Ø4 tube fitting instrument air inlet
Instrumentation air
Sensor headinjection
9

Installation
screw

8.5

7.5

24

33

Ø22

48

Measurement distance

Water cooling hood(option)

8

B

42

A

42
Cooling water inlet
for tube 6×4
0.5MPa Max

Mounting screw
4-M3 depth 4

Mounting bracket

TMAX-B (option)
Model

TMWX-A1
TMWX-A2
TMWX-A3

A
62

B
35

Accessories

Hood Model

98

50

TMHX-CLE

TMPX-15W

86

50

Accessory : C

for tube 6×4※not included : TMHX-CLE

Ø34

for tube 6×4
0.5MPa Max
Cooling water outlet
for tube 6×4

Sensor head

33

9Cooling water inlet

64

Ø4 tube fitting
instrument air inlet

Nut

Measurement distance

80

9

42

7.5

32

TMPX-10W
TMPX-15W

A
B※

TMWX-B4

Use with the sensor head to be installed where the ambient
temperature is above 50℃.

Water cooling hood
TMPX-25W (option)

Cooling water outlet

Mounting bracket
TMAX-B (option)

11/12

Screw 4-M3 depth4

8.5

Water cooling jacket*Image

10.2

20

6.5

11.7

26 ±0.5

1632 ±0.5

52 ±0.533

40 ±0.5
38.4 ±0.5
5.5

31

Ø
21
ho
le

13

Mounting holeSensor head
2-Ø6.2hole

Weight : 100g

Mounting holeSensor head
2-Ø6.2hole

Weight : 65g

Accessory : C

Use with the sensor head toprotect from electromagnetic
at applications such as highfrequency heating.

56 ±0.5

Use with the sensor head to protectfrom electromagnetic atapplications such as highfrequency heating.

TMSX-B4

Shield case54.4 ±0.5

Shield case48

Air controller for the container
purge air inlet dried air for 4×2.5 tube
(for avoiding condensation)

Infrared Thermometer TMHX-C series Specifications

Accessories
Extension cable
Extension cable : 5m
e-con female / e-con male

TMBX-E05

Branch cable

Accessory : A,B,C

For setting unit when thermometeris used alone. e-con branch

Weight : 390g

Resistor

TR-251N / TR-501N

Noise suppressor for analog.

Weight : 0.5g

Ferrite core
For power noise. Install to cables.

Accessory : A,B,C

Weight : 85g

Capacitor

Analog output
Accessory : A,B,C
TR-251N : 0 〜 5V
TR-501N : 0 〜 10V
To change from current to voltage : 0 〜 20mA

TMBX-B01

TC-105N
Accessory : A,B,C

Weight : 0.5g

FC-2032
Accessory : A,B,C

Weight : 23g●The contents described in this catalog are subject to change for improvement without notice.

●If you have any questions, contact us at the address orlinks shown below.

http://www.japansensor.co.jp
info@japansensor.co.jp

12/12

TMHX-Spec-C-V201906P201906 1K