SDG7000A Series
Arbitrary Waveform
Generator

Data Sheet
EN01C

Key Features
Dual channel differential/single-ended output, 16-bit
LVDS/LVTTL digital bus output.
High-performance sampling system with 5GSa/s sample rate

SDG7102A

and 14 -bit vertical resolution.
1 GHz maximum bandwidth.

SDG7052A

Generates arbitrary waveform with sample rates of 0.01 Sa/s
~ 2.5 GSa/s, with maximum memory depth of 512 Mpts, andSDG7032A

provides segment editing /playback functions.
Generates vector signals with up to 500 MS/s symbol rate.
Generates low jitter pulses with 1 ns minimum pulse widthand 500ps minimum edge.
Up to 1 GHz bandwidth White Gaussian Noise and the
bandwidth is adjustable.

Product Overview

Supports PRBS up to 625 Mbps.

SIGLENT’s SDG7000A is a family of dual-channel Arbitrary

The digital bus can output digital signals up to 1 Gbps.

Waveform Generators that feature up to 1 GHz bandwidth,

Supports frequency hopping output under sine and AM

a maximum sample rate of 5 GSa/s and 14-bit verticalmodulation, providing output list frequency or PRBS random

resolution. It can generate arbitrary waveforms point bypoint with a maximum 2.5 GSa/s sample rate and vector

output within a specified range of frequencies
Supports multi pulse output function
Supports analog/digital modulation, sweeping and bursting.

signal with a maximum 500 MS/s. It also has the ability toEnhanced dual channel operation functions: inter channel

generate a variety of signals such as continuous wave,

tracking, coupling and copying; Dual channel superposition

Pulse, Noise, PRBS patterns, and a 16-bit digital bus. It

function; Supports mutual modulation between channels.

supports the generation of complex signals such as

The 24 Vpp analog output is superimposed with ± 12 Vdc

modulation,offset to provide a maximum output range of ± 24 V (48 V).

sweeping,

bursting

and

dual

channelcopying/coupling/tracking and superposition. The output

High precision Frequency Counterare differential/single-ended and support a maximum

5-inch capacitive touch screen with resolution of 800x480;

output range of ± 24 V. The instrument can ensure a large

Supports external mouse and keyboard operation; Supports

amplitude under high-frequency which eliminates an

WebServer to control the instruments remotely.

external power amplifier in some applications andSupports multiple interfaces: 10MHz In, 10MHz Out, Trigger

addresses a wider range of requirements.

In/Out, Markers etc.
Supports SCPI command for easy integration into test
systems.

int.siglent.com

1

SDG7000A Arbitrary Waveform Generator

Models and Key Specifications
Model

SDG7102A

Number of channels

2 Differential/Single-ended

Bandwidth

1 GHz

Sample rate

5 GSa/s

Vertical resolution

14-bitArbitrary waveform

0.01 Sa/s ~ 2.5 GSa/s sample rate; 24 pts ~ 512 Mpts/ch memory depth, with segment
editing and playback

Vector signal (Optional)

500 MS/s max symbol rate; Carrier DC ~ 1 GHz settable. Includes modulation modes
such as ASK, PSK, FSK and QAM. EasyIQ software provides vector signal creation andediting; SigIQPro software provides waveform generation capabilities for Bluetooth's BR,
EDR, BLE, IoT's 802.15.4 O-QPSK, SUN FSK, SUN OFDM, ITU-T G.9959, Custom
OFDM, and other protocol waveforms

Continuous waveform

Up to 1GHz, supports harmonic generation function

Pulse

Min pulse width 1 ns, min. edge 500 ps pulse with low jitter, the rise/fall edge isindependently fine adjustable, and the pulse width is fine adjustable

Noise

Bandwidth 1 mHz ~ 1 GHz adjustable

PRBS

Bit rate 2 μbps ~ 625 Mbps, length PRBS3 ~ PRBS32

Complex signal generation

Supports internal/external modulation, AM, FM, PM, PWM, FSK, PSK, ASK, etc.;
Supports sweep; Support burst

Dual-channel function

Inter channel tracking, coupling, and copying. Dual channel superposition function.
Supports mutual modulation between channels

Output range

24 Vpp analog output superimposed ± 12 V DC offset, supports a maximum output rangeof ± 24 V (48 V)

Digital bus(Optional)

SDG7052A

SDG7032A

500 MHz

350 MHz

16-bit, LVTTL or LVDS output
Bit rate: 1 μbps ~ 1 Gbps
USB 2.0 Host x3, USB 2.0 Device(USBTMC)Interface

LAN 10M/100M (VXI-11/Telnet/Socket/WebServer)
EXT MOD/CNT, 10MHz In, 10MHz Out, Marker x2, Trigger In/Out
5” TFT-LCD with capacitive touch screen (800x480)

Interaction

Supports mouse operation
Supports Webserver
Supports SCPI control

2

int.siglent.com

SDG7000A Arbitrary Waveform Generator

Characteristics
Multi-functional Waveform Generator

The SDG7000A series integrates multiple waveform generator functions from DC to continuous waves up to 1
GHz, which can replace RF signal generators in some applications. It adopts Siglent’s TrueArb point-by-point
arbitrary waveform generation technology, which enables user-adjustable output sample rates from 0.01 Sa/s
to 2.5 GSa/s with excellent jitter performance and the generation of I/Q vector signals with a maximum
settable bandwidth greater than 500 MHz. Using the benefits of Siglent’s EasyPulse architecture, a low jitter
pulse with a minimum pulse width of 1 ns can be generated. The SDG7000A also features a Gaussian noise
output with adjustable bandwidth, Pseudo-random code generation, an optional 16 channels of digital signal
output for synthesizing digital communications, and much more.

Wide Range Amplitude Output

24Vpp analog output superimposed with ± 12 Vdc
offset, providing a maximum output range of ± 24 V
(48 V).

int.siglent.com

3

SDG7000A Arbitrary Waveform Generator

Excellent Arbitrary Waveform Generation
AFG mode uses traditional DDS technology to generate arbitrary waveforms
AWG mode uses the innovative TrueArb technology, with an adjustable sample rate from 0.01 Sa/s~ 2.5 GSa/s and jitterless than 20 ps. It not only has all the advantages of traditional DDS technology, but also overcomes its intrinsic jitter anddistortion defects. The flexible platform also provides zero order hold, linear and sinc interpolation methods for increased
flexibility when creating complex waveforms.

Sequence editing and playback The SDG7000A supports
up to 1024 arbitrary wave segments, each of which can be
set with a maximum of 65535 repetitions. When switching
between segments, the output seamlessly moves from thelast point of the previous segment to the first point of the next
segment without generating an idle level. It is suitable forapplications with high requirements for waveform switching.

EasyWaveX supports extensive arbitrary wave editing
functions including manual, linear, coordinate, and equation
drawing that facilitate rapid generation of the required
waveforms. The EasyWaveX editing software is embeddedin the SDG7000A, and can also be installed in a computer,interacting with the SDG7000A over USB or LAN interfaces.

High-Speed Low Jitter Pulse

Low jitter When a Square/Pulse waveform is generated by traditional DDS, there can be additional jitter if thesample rate is not an integer-related multiple of the output frequency. EasyPulse technology successfully
overcomes this weakness in DDS designs and helps to produce low jitter Square/Pulse waveforms.

4

int.siglent.com

SDG7000A Arbitrary Waveform Generator

High speed The minimum 1 ns pulse width, can begenerated at any frequency. The pulse width can be finely
adjusted in steps of 10 ps.

Flexible edge Adjustable fine step resolution to 100 ps. The
minimum edge is 500 ps and can be generated at any
frequency. The rising/ falling edge can be set respectively
and can be used to generate asymmetric pulse.

Vector Signal Output (Optional)

The SDG7000A can generate common modulation types of IQ signals, such as ASK, FSK, PSK, QAM. With the innovative
resampling technology, excellent EVM performance can be obtained at any symbol rate between 250 S/s ~ 500 MS/s. The
built-in digital quadrature modulator can modulate the carrier of the IQ signal to any frequency between 0 Hz~1 GHz. The
EasyIQ software can be used to generate and edit various types of IQ signals.

int.siglent.com

5

SDG7000A Arbitrary Waveform Generator

Complex Signal Generator

Modulation A variety of analog and digital modulation
modes such as AM, FM, PM, FSK, ASK, PSK,DSB-SC, and PWM are supported. There are threemodulation sources: Internal, External, and Channel.

Sweep and Burst Sweep have three types of“Frequency”, ”Amplitude” and ”Freq&Ampl”. Sweep
supports "Line", "Log" and “Step” modes, while Burstenables "NCycle" and "Gated" modes. Both Sweep
and Burst support trigger sources: Internal, External,and Manual.

Harmonics Function provides the ability to add higher-order elements to your signal.

16 Channel Digital Output (Optional)

Purchase the corresponding digital bus kit to get 16-channel LVTTL or LVDS output with a bit rate of 1 μbps ~ 1 Gbps.
Combine the digital bus with the analog channels to realize mixed-signal outputs.

6

int.siglent.com

SDG7000A Arbitrary Waveform Generator

Enhanced Dual Channel Functionality
Two Dual-Channel Operation Mode

Independent mode enables the two channels to be used as two independent generators.
Independent mode also eliminated the discontinuity on the output when parameters (frequency,
amplitude) change.

Phase-Locked mode Automatically aligns the phases of each output.

Track/Copy/Coupling The track, copy and coupling functions
between the two channels can quickly transfer the parametersof one channel to the other according to the requirements,
greatly simplify the operation and meet the requirements offast and synchronous switching waveforms.

Waveform Combining Superimposes CH1 and CH2
waveforms internally and provides the combined waveform toa user-selected output. It easily combines basic waveforms,
random noise, modulation signals, sweep signals, burst
signals, EasyPulse waveforms, and TrueArb waveforms
without external connections or complex editing.

Channel Modulation One channel can modulate the otherwithout external connections. This feature provides an easy
method for complex modulation waveform creation. The
modulating wave channel can be directly output andcompared with the modulated signal.

int.siglent.com

7

SDG7000A Arbitrary Waveform Generator

Hopping Supports three types of frequency hopping output modes: list mode, random mode, and random list mode.

Multi pulse Supports multi pulse output. The positive and negative pulse widths of each pulse can be individually edited andset through EasywaveX tools.

8

int.siglent.com

SDG7000A Arbitrary Waveform Generator

Specifications
Unless otherwise specified, all specifications can be guaranteed to meet the following conditions:• Within the validity period of product calibration.
• Within the ambient temperature range of 18 ℃ ~ 28 ℃.
• The instrument is powered on and operating for more than 30 minutes.

Frequency
Parameter

Min.

Resolution

1μ

Typ.

Max.

Unit

Condition & Note

Hz

Standard time base
-1

+1

ppm

25 ℃

-2

+2

ppm

0 ~ 50 ℃

1st-year aging

-0.5

+0.5

ppm

20-year aging

-3.0

+3.0

ppm

Initial accuracy

OCXO option
Frequency

10M

Hz

Initial accuracy

-100

+100

ppb

25 ℃

Temperature stability

-1

+1

ppb

0 ~ 50 ℃

1st-year aging

-50

+50

ppb

Max.

Unit

Condition & Note

1μ

1G

Hz

SDG7102A

1μ

500M

Hz

SDG7052A

1μ

350M

Hz

SDG7032A

Harmonic distortion

-55

dBc

≤500MHz, 0 dBm(Single-ended）

-40

dBc

＞500MHz, 0 dBm

Total Harmonic

-55

dBc

≤500MHz, 0 dBm(Differential)

-45

dBc

＞500MHz, 0 dBm

Non-harmonic spurious(Single-ended)

-65

dBc

0 dBm

Non-harmonic spurious(Differential)

-60

dBc

0 dBm

dBc/Hz10 MHz@10 kHz offset, 0 dBm

Sine
Parameter

Frequency

Phase noise

Min.

Typ.

-138

Custom harmonic number
Custom harmonic typeint.siglent.com

16

order

Even, Odd, All

9

SDG7000A Arbitrary Waveform Generator

Arbitrary Wave
Parameter

Min.

Typ.

Max.

Unit

2.5 G

Sa/s

Condition & Note

AWG Mode
Sample rate

Waveform length

0.01

24

512M

Vertical resolution

14

JitterInterpolation mode

50

pts

In sequence mode, when thesegments＞1. The minimum
waveform length is 64. When thelength is less than 256 points, it must
be an integer multiple of 16 points

bit

The data storage format is 16-bit

psCycle to Cycle RMS value , ”010101”
pattern, 1 Vpp,50 Ω load, 2.5 GSa/S

0-order hold, linear, sinc, sinc27, sinc13
Up to 4096 segments. Each segment can be repeated for
1~65535.

Sequence

Run mode: Continuous, Single/Burst, Infinite, Stepped,
Advanced
Trigger source: Button, Timer, External

SourceBuild-In, From File, EasyWaveX

AFG Mode
Sample rate

2.5 G

Sa/s

Waveform length

32 k

pts

Frequency range

1μ

100 M

Hz

SDG7102A

1μ

50 M

Hz

SDG7052A

1μ

35 M

Hz

SDG7032A

450

ps

Rise/Fall time

300

345

SourceBuild-In, From File, EasyWaveX

10% ~ 90%, 1 Vpp step signal,
50 Ω load

Vector (I/Q) signal (optional)
Parameter

Min.

Symbol rate

250

Waveform length

Typ.

Max.

Unit

Condition & Note

500 M

Symbol/s

Limited by oversampling multiple

512 M

pts

1G

Hz

Limited by the bandwidth parameterbit

The data storage format is16-bit

Carrier frequency

0

Vertical resolution

14

Modulation type2ASK, 4ASK, 8ASK, BPSK, QPSK, 8PSK, DBPSK, DQPSK,OQPSK, D8PSK,8QAM, 16QAM, 32QAM, 64QAM,
128QAM, 256QAM, 2FSK, 4FSK,8FSK, 16FSK, MSK,
MultiTone, Custom

Supported by EasyIQ

150

MHz

Calibrated and tested specifications

Modulation
bandwidth

625

MHz

Determined by the symbol rate andfilter settings

EVM

2.5

%

64QAM, Single-ended, thebandwidth is 150MHz

10

int.siglent.com

SDG7000A Arbitrary Waveform Generator

0.8%

64QAM, Single-ended, 0.3 Vrms, thebandwidth is 100 MHz,

0.5%

64QAM, Single-ended, 0.3 Vrms, thebandwidth is 50 MHz,

0.3%

64QAM, Single-ended, 0.3 Vrms, thebandwidth is 10 MHz,

Data type

PN7, PN9, PN15, PN23, User files, Custom constellation

SourceBuilt-In, external file, EasyIQ

IQ compensation

Gain Balance, Offset, Angle

Supported by EasyIQ software

Pulse
Parameter

Frequency

Pulse width

Min.

Typ.

Max.

Unit

Condition & Note

1μ312.5 MHz

SDG7102A

1μ

150 M

Hz

SDG7052A

1μ

100 M

Hz

SDG7032A

ns

10 ps resolution. The maximum pulse
width is limited by the frequency setting,and the minimum pulse width isindependent of the frequency

1± (0.01% +0.3ns)

Pulse Width accuracy

Rise time(specified range )Rise time(setting range)

Fall time(specified range)

Fall time(setting range)

1n

75

s

Amplitude ≤ 3Vpp , | offset | ≤ 3V, 10% ~90%, 100 ps resolution. Specifications
such as overshoot, jitter, output range, andpulse width accuracy can only be met
within this range. The minimum value thatcan be set is limited by the output
amplitude.

0.5 n

75

s

10% ~ 90%, 100 ps resolution. The
minimum value that can be set is limited by
the output amplitude.

1n

75

s

Amplitude ≤ 3Vpp , | offset | ≤ 3V, 10% ~90%, 100 ps resolution. Specifications
such as overshoot, jitter, output range, andpulse width accuracy can only be met
within this range. The minimum value thatcan be set is limited by the output
amplitude.

0.5 n

75

s

10% ~ 90%, 100 ps resolution. The
minimum value that can be set is limited by
the output amplitude.

3%

100 kHz, 1 Vpp, 50 Ω load, 1 ns edge

99.999%0.001% resolution. Limited by frequency
setting20

ps

Cycle to cycle rms. >10 kHz, 1 Vpp, 50 Ω
load

OvershootDuty cycle
Jitterint.siglent.com

0.001
10

11

SDG7000A Arbitrary Waveform Generator

SquareParameter

Frequency

Rise /fall times

Min.

Max.

Unit

Condition & Note

1μ

240 M

Hz

SDG7102A

1μ

150 M

Hz

SDG7052A

1μ

100 M

Hz

SDG7032A

1.1

ns

10% ~ 90%, 1 Vpp, 50 Ω load. It varies
with the output amplitude.

3%

100 kHz, 1 Vpp, 50 Ω load, 1 ns edge

99.999%0.001% resolution. Limited by frequency
setting

10

20

ps

cycle to cycle rms, >10 kHz,1Vpp, 50 Ω
load

Typ.

Max.

Unit

Condition & Note

1m

1G

Hz

SDG7102A

1m

500 M

Hz

SDG7052A

1m

350 M

Hz

SDG7032A

Max.

Unit

Condition & Note

2μ

625 M

bps

SDG7102A

2μ

625 M

bps

SDG7052A

2μ

200 M

bps

SDG7032A

1μs

10% ~ 90%, 1 Vpp, 50 Ω load

Max.

Unit

Condition & Note

0.85

Typ.

1.0

OvershootDuty cycle

0.001

JitterNoiseParameter

Bandwidth (-3dB)

Min.

PRBS
Parameter

Bit rate

Min.

Typ.

Pattern length

2m-1, m = 3,4,…,32

Rise/fall time

0.5 n

Ramp
Parameter

Min.

Typ.

Frequency

1μ

10 M

Hz

Symmetry

0

100

%

1.5

%

Linearity

1% resolution
Percentage of peak output, 1 kHz,
1 Vpp, 50% symmetry

DC
Parameter

Min.

Accuracy

Typ.

Max.

Unit

± (1% + 2mV)

Condition & Note
HiZ load

-6

+6

V

50 Ω load

-12

+12

V

HiZ load

Output Range

12

int.siglent.com

SDG7000A Arbitrary Waveform Generator

OutputParameter

Min.

Typ.

Max.

Unit

Condition & Note

+12V

HiZ load, divide by 2 at 50 Ω loaddB

50 Ω load, 0.32 Vpp, compare to1 MHz Sine

Single-ended
Offset-12± (1% + 2
mV)

Offset accuracy
Amplitude flatness

-0.3

+0.3
± (1% + 1
mVpp)

Amplitude accuracy

10 kHz sine, 0 V offset

1m

12

Vpp

≤ 40MHz, 0 V offset

1m

6

Vpp

40 MHz ~120 MHz (including 120 MHz)

1m

3

Vpp

Arb output range

1m

12

Vpp

(50Ω load)*

1m

6

Vpp

1m

12

Vpp

> 120 MHz≤ 20 MHz, 0 V offset> 20 MHz≤ 20 MHz, 1 ns edge, ≥ 10 ns pulse width

1m

6

Vpp

20 MHz ~ 60 MHz (including 60 MHz),
1 ns edge, ≥ 10 ns pulse width

1m

3

Vpp

> 60 MHz, 1 ns edge, ≥ 10 ns width pulse

Sine output range(50Ω load) *Pulse/square output
range(50Ω load)*

Noise output range(50Ω load)*

PRBS output range(50Ω load)*

1m

0.669

Vrms

Mean = 0; Close the bandwidth setting.
≤ l67 mVrms, upper limit of noise bandwidth
1 GHz>167 mVrms, lower noise bandwidth limit

1m

12

Vpp

≤ 40 Mbps, 0 V offset

1m

6

Vpp

40 Mbps ~ 120 Mbps (including 120 Mbps)> 120 Mbps

1m

3

Vpp

Ramp output range(50Ω load)*

1m

12

Vpp

Internal resistance

49

51

Ω

Current output

-240

+240

mA

Protection

Over-current protection,
Over voltage protection

Crosstalk

-60

dBc

CH1=CH2=0 dBm, Sine, 50 Ω load

Skew

20

ps

The same amplitude setting of two channels

50

100 kHz sine wave

Differential
Amplitude flatness

-0.3

+0.3

dB

100Ω load , 0.5 Vpp, compare to 1 MHz Sine

Output

20 m

2

Vpp

Differential peak to peak, 100 Ω differentialload, common offset = 0 V

Offset-1+1

V

Differential offset, 100 Ω differential loadCommon mode

-1

+1

V

Load = HiZ

Protection

Over voltage protection

Crosstalk

-60

dBc

CH1=CH2=0 dBm, Sine, 50 Ω load

Skew

20

ps

The same amplitude setting of two channels

Other output characteristics
Output polarity

Normal, Invert

Noise superposition

The minimum SNR can be set to 0.1 dBc

Digital filter

100

BW

MHz

BW is the maximum output frequency*Note: The specification will be multiplied by 2 while applied to a HiZ load.

int.siglent.com

13

SDG7000A Arbitrary Waveform Generator

Modulation
Parameter

Min.

Typ.

Max.

Unit

Condition & Note

AM
CarrierSine, Square, Ramp, Arb

Modulation source

Internal, External, Channel

Modulation wave

Sine, Square, Ramp, Noise, Arb

Modulation depth

0

120

%

Modulation
frequency

1m

2M

Hz

Source = Internal

Source = Internal

FM
CarrierSine, Square, Ramp, Arb

Modulation source

Internal, External, Channel

Modulation wave

Sine, Square, Ramp, Noise, Arb

Frequency deviation

0

0.5 x BWHz

BW is the max. frequency. Limited byfrequency setting

Modulation
frequency

1m

2M

Hz

Source = Internal

Source = InternalPM
CarrierSine, Square, Ramp, Arb

Modulation source

Internal, External, Channel

Modulation wave

Sine, Square, Ramp, Noise, Arb

Phase deviation

0

360

°

Modulation
frequency

1m

2M

Hz

Source = Internal

Source = Internal

PWM
Carrier

Pulse

Modulation source

Internal, External, Channel

Modulation wave

Sine, Square, Ramp, Noise, Arb

Modulation
frequency

1m

2M

Source = Internal
Hz

Source = Internal

ASK
CarrierSine, Square, Ramp, Arb

Modulation source

Internal, External, Channel

Modulation wave

Square with 50% duty cycle

Keying frequency

1m

2M

Source = Internal
Hz

Source = Internal

FSK
CarrierSine, Square, Ramp, Arb

Modulation source

Internal, External, Channel

Modulation wave

Square with 50% duty cycle

Keying frequency

1m

2M

Source = Internal
Hz

Source = Internal

PSK
CarrierSine, Square, Ramp, Arb

Modulation source

Internal, External, Channel

Modulation wave

Square with 50% duty cycle

Keying frequency

1m

14

2M

Source = Internal
Hz

Source = Internal

int.siglent.com

SDG7000A Arbitrary Waveform Generator

Burst
Parameter

Min.

Typ.

Max.

Unit

CarrierSine, Square, Ramp, Pulse, Noise, Arb

Type

Count (1-1000000 periods), Infinite, Gated

Cycles1

10^13periods

Burst Period

1μ

900k

s

Carrier frequency

2m

BW

Hz

Start/Stop phase

0

360

°

Trigger source

Internal, External, Manual

Gated source

Internal, External

Trigger delay

2.079 μ

10

s

Max.

Unit

Condition & Note

Source =External, Manual

BW is the max. output frequency

1 kHz Pulse. The min. delay is limited by
waveform and frequency

Sweep
Parameter

Min.

Typ.

CarrierSine, Square, Ramp, Arb

Type

Frequency, Amplitude, Freq&Ampl

Mode

Linear, Logarithmic, Step

Condition & Note

Linear: Up, Down, Up & Down
Direction

Logarithmic: Up, Down
Step: Up, Down, Up & Down

Sweep time

10u

1000

s

Carrier frequency

1μ

BW

Hz

Trigger source

Internal, External, Manual

Trigger delay

1.963 μ

10

s

Max.

Unit

BW is the max. output frequency

CounterParameter

Min.

Typ.

Mode

Totalizer, Frequency

Frequency
Parameter

Frequency, Period, Positive/Negative Pulse Width,
Duty Cycle

Coupling mode

AC, DC, FREJ

Condition & Note

100 m

400 M

Hz

DC coupling

1

400 M

Hz

AC coupling

100 mVrms±2.5 V

DC coupling < 100 MHz

200 mVrms±2.5 V

DC coupling, 100 MHz ~ 200 MHz

500 mVrms±2.5 V

DC coupling, > 200 MHz

100 mVrms

5 Vpp

AC coupling, < 100 MHz

200 mVrms

5 Vpp

AC coupling, 100 MHz ~ 200 MHz

500 mVrms

5 Vpp

AC coupling, > 200 MHz

Frequency range

Input amplitude

Input impedance

int.siglent.com

1M

Ω

15

SDG7000A Arbitrary Waveform Generator

Digital (optional)
Parameter
Bit rate

Min.

Max.

Unit

Condition & Note

1m

Typ.

1G

bps

LVDS

1m

200 M

bps

LVTTL

Condition & Note

InterfaceParameter

Min.

Typ.

Max.

Unit

Frequency

9.999910.0000

10.0001

MHz

Amplitude

1.4

10MHzInput

Input impedance

Vpp
5

kΩ

10.000

MHz

3.3

Vpp

50

Ω

10MHz Output
Frequency
Amplitude2

Output impedance

Output sine wave

Modulation Input
Frequency

0

1M

Hz

Input impedance

10kΩ

Amplitude@100%modulation
depth

±5

Vpp

Trigger Input
VIH

2

5.5

V

VIL

-0.5

0.8

V

Input impedance
Pulse width

kΩ

100
100

ns

Response time

2.28

μs

Sweep

1.96

μs

Burst, non Pulse/Square

2.07

μs

Burst mode, Pulse/SquareV

IOH = 8 mA

V

IOL = -8 mA

Trigger output
VOH

3.8

VOL
Output impedance

0.44

Ω

50

Frequency

1M

Hz

Marker Output
Frequency

3.3

Vpp

Pulse width

108

ns

Rise/fall edge

25.6

ns

Output impedance50ΩJitter400

ps

Trigger delay

3.2

μs

16

Cycle to cycle rms

int.siglent.com

SDG7000A Arbitrary Waveform Generator

GeneralParameter

Min.

Typ.

Max.

Unit

Condition & Note

Power
Voltage

100 - 240 Vrms (± 10%), 50/60 Hz

Power consumption

90

110

W

Touch screen
Dimensions

5.0

inch

Resolution

800 x 480

pixel

Color depth

24

Contrast Ratio

500

600

Luminance

200

250

Touch Screen Typecapacitive

bit

cd/m2

Environment
Operatingtemperature

0

50

℃

Non-operating
temperature

-20

60

℃

5

90

%RH

30 ℃

5

50

%RH

50 ℃

5

95

%RH

Operating altitude

3048

m

Non–operating
altitude12192

mOperating humidity
Non-operating
humidity

254

Meets EMC directive (2014/30/EU), meets or exceeds IEC 61326-1:2012/EN61326-1:2013 (Basic)

EMC

Conducted disturbance

CISPR 11/EN 55011

CLASS A group 1,
150 kHz-30 MHz

Radiated disturbance

CISPR 11/EN 55011

CLASS A group 1,
30 MHz-1 GHz

Electrostatic discharge (ESD)

IEC 61000-4-2/EN 61000-4-2

4.0 kV (Contact), 8.0 kV (Air)Radio-frequency
electromagnetic field Immunity

IEC 61000-4-3/EN 61000-4-3

10 V/m (80 MHz to 1 GHz)
3 V/m 00(1.4 GHz to 2 GHz)
1 V/m (2.0 GHz to 2.7 GHz)Electrical fast transients (EFT)IEC 61000-4-4/EN 61000-4-4

2 kV (Input AC Power Ports)

SurgesIEC 61000-4-5/EN 61000-4-5

1 kV（Line to line）
2 kV（Line to ground）

Radio-frequency continuous
conducted Immunity

IEC 61000-4-6/EN 61000-4-6

3 V, 0.15-80 MHz

IEC 61000-4-11/EN 61000-4-11

Voltage Dips:
0% UT during 1 cycle;
40% UT during 10/12 cycles;
70% UT during 25/30 cycles
Voltage interruptions:
0% UT during 250/300 cycles

Voltage dips and interruptions

SafetyUL 61010-1:2012/R: 2018-11; CAN/CSA-C22.2 No. 61010-1:2012/A1:2018-11.

Mechanical
Dimensions

338 × 113 × 369

mm

Net weight

4.4

kg

int.siglent.com

W×H×D

17

SDG7000A Arbitrary Waveform Generator

Ordering Information
Product Model

Description

SDG7102A

1 GHz, 5 GSa/s, 14-bit, 512 Mpts, 5-inch capacitive touch screen

SDG7052A

500 MHz, 5 GSa/s, 14-bit, 512 Mpts, 5-inch capacitive touch screen

SDG7032A

350 MHz, 5 GSa/s, 14-bit, 512 Mpts, 5-inch capacitive touch screenStandard Configurations

Quantity

USB cable

1

BNCcoaxial cable2

Quick start1

Power cord1

Wireless mouse1

Optional Configurations

Model

20 dB Attenuator

ATT-20dB

Single Instrument Rack Mount Kit

SSG-RMK

USB-GPIB Adapter

USB-GPIBHigh precision OCXO (Installed at the factory, cannot be added after purchase)

10M_OCXO_L

Digital Bus Kit-LVTTL

DIG-LVTTL

Digital Bus Kit-LVDS (Without RF cables)

DIG-LVDS

Digital Bus Kit-LVDS (With 32 RF cables)

DIG-LVDS-2

IQ Signal Generator Function (software)

SDG-7000A-IQ

350 MHz to 500 MHz bandwidth upgrade (software）

SDG-7000A-BW05

500 MHz to 1 GHz bandwidth upgrade (software）

SDG-7000A-BW10

18

int.siglent.com

SDG7000A Arbitrary Waveform Generator

About SIGLENTSIGLENT is an international high-tech company, concentrating on R&D, sales,
production and services of electronic test & measurement instruments.

SIGLENT first began developing digital oscilloscopes independently in 2002.
After more than a decade of continuous development, SIGLENT has
extended its product line to include digital oscilloscopes, isolated handheld
oscilloscopes,
generators,

function/arbitrary
spectrumwaveform

analyzers,

vectorgenerators,
network

RF/MWsignalanalyzers,

digitalmultimeters, DC power supplies, electronic loads and other general purpose
test instrumentation. Since its first oscilloscope was launched in 2005,
SIGLENT has become the fastest growing manufacturer of digital
oscilloscopes. We firmly believe that today SIGLENT is the best value inelectronic test & measurement.

Headquarters:
SIGLENT Technologies Co., Ltd
Add: Bldg No.4 & No.5, Antongda Industrial
Zone, 3rd Liuxian Road, Bao'an District,
Shenzhen, 518101, China
Tel: + 86 755 3688 7876
Fax: + 86 755 3359 1582
Email: sales@siglent.com
Website: int.siglent.com

North America:
SIGLENT Technologies America, Inc
6557 Cochran Rd Solon, Ohio 44139
Tel: 440-398-5800
Toll Free: 877-515-5551
Fax: 440-399-1211
Email: info@siglentna.com
Website: www.siglentna.com

Europe:Follow us on
Facebook: SiglentTech

SIGLENT Technologies Germany GmbH
Add: Staetzlinger Str. 70
86165 Augsburg, Germany
Tel: +49(0)-821-666 0 111 0
Fax: +49(0)-821-666 0 111 22
Email: info-eu@siglent.com
Website: www.siglenteu.com

int.siglent.com

19