X-Scan c5 SSeries
X-ray linear array detector

X-Scan c5 Series is an enhanced product family of fast andnd compact
linear array detectors for Food, Pharmaceutical and Industrial
strial inspection
applications. In order to cut down the overall system costt and to simplify
the structure of industrial X-ray machines, the X-Scan c5 not only
introduces a broad range of new features, but also significantly
ficantly improved
radiation hardness for the detector lifetime and reduced dark image noise toachieve better quality.
With more and more demanding market requirements, the X-Scan c5 can also be
cost-optimized for the X-ray machines targeted to replacing
cing metal detectors in the foodprocessing lines.

Features:xxxxxxxxxxxx

Benefits:

Detectable X-ray energy range: 20KeV - 160KeV
Active length: from 102 mm to 614 mm
Pixel size options 0.2 mm, 0.4 mm and 0.8 mm
Conveyor speed: up to 3.3 m/s
Up to 14-bit AD, Dynamic range: > 4000
Firmware upgrade via USB 2.0 interface
Command set compatible with f3 and c4 series productsCompact, water resistance enclosure with IP67 classification
High tolerance (up to 6K volts contact discharge) to electrostatic discharge (ESD)
USB interface support for multi-view systems (up to 4pcs detector to single PC)
Enhanced features including binning function and temperature drift correction
Significantly improved radiation hardness

xxxxxxxxx

Versatile in Food, Pharmaceutical and Industrial inspection applications
Simplify X-ray system design in order to cut down overall system cost
Radiation hardness improvement for detector lifetime
Robust design for harsh environment
Industry leading sensitivity and speed
Optimized design for food and other fast low energy applications
Cost-effective solution with high performance DT proprietary
photodiode and ASIC designs
Software with USB 2.0 interface as easy plug-and-play system
Easy software design based on DT control library and sample codes

Enclosures of the x-scan c5 Series
Enclosure Model

Active length

LengthWidth

HeightMax Weight

X-Scan 0.xc5-102

102 mm (4”)

230 mm (9.1”)

200 mm (7.9”)30 mm (1.2”)

2.5 kg (5.5 lbs)

X-Scan 0.xc5-154

154 mm (6”)

230 mm (9.1”)

200 mm (7.9”)30 mm (1.2”)

2.5 kg (5.5 lbs)

X-Scan 0.xc5-205

205 mm (8.1”)

230 mm (9.1”)

200 mm (7.9”)30 mm (1.2”)

2.5 kg (5.5 lbs)

X-Scan 0.xc5-256

256 mm (10.1”)

355 mm (13.2”)

200 mm (7.9”)30 mm (1.2”)

3.1 kg (6.8 lbs)

X-Scan 0.xc5-307

307 mm (12.1”)

335 mm (13.2”)

200 mm (7.9”)30 mm (1.2”)

3.7 kg (8.2 lbs)

X-Scan 0.xc5-410

410 mm (16.1”)

438 mm (17.2”)

200 mm (7.9”)30 mm (1.2”)

4.8 kg (10.6 lbs)

X-Scan 0.xc5-512

512 mm (20.2”)

540 mm (21.3”)

200 mm (7.9”)30 mm (1.2”)

5.9 kg (13.0 lbs)

X-Scan 0.xc5-614

614 mm (24.2”)

643 mm (25.3”)

200 mm (7.9”)30 mm (1.2”)

7.0 kg (15 lbs)

Detection Technology, Inc. | Piisilta 1, 91100 Ii, FINLAND | Tel. +358 8 553 6600 | Fax +358 8 553 6611 | http://www.deetee.com | contact@deetee.com

DS0000076A

X-Scan c5 Series

General Characteristics

X-ray linear array detector

Product

X-Scan 0.2c5

X-ray tube voltage Vp Range

X-Scan 0.4 c5

X-Scan 0.8c5

Notes

40-160 kVp

Scintillator material

GOS

Active area lengths

Note 1) Dynamic range is defined as Saturation
signal/RMS Noise, with the default gain setting (feedback capacitance 1pf) of ASIC. Available gain options
are listed below:

102-614 mm

Number of pixels

512-3072

256-1536

128-768

Pixel pitch (spacing)

0.2 mm

0.4 mm

0.8 mmPixel height

0.3 mm

0.6 mm

0.8 mmPixel width

0.15 mm0.32 mm0.72 mm

Maximum scanning speed- Active lengths of (102-307 mm)- Active lengths of (410-614 mm)

60 cm/s32 cm/s

167 cm/s
121 cm/s

333 cm/s
333 cm/s

Minimum integration time
- Active lengths of (102-307 mm)- Active lengths of (410-614 mm)

0.33 ms
0.62 ms

0.24 ms
0.33 ms

0.24 ms
0.24 msGS001

Cf [pf]

10,5Note 2) Digital interface uses 16-bit per pixel, butuser can select actual output data to be 16,14,12,10or 8 bits. The default value is 16-bits.
Note 3) Non-Linearity is defined as max deviationfrom ideal response line defined by zero offset andsignals at 80% of dynamic range.
Note 4) The Operational voltage to be supplied tomodule. External power supply for 100-240 AC isprovided if requested.

Maximum integration time

128 ms

A/D resolution

14 bits1)

Dynamic range> 40001)

Note 5) Not corrosive environment

Data digital interface

16 bits2)

InterfaceUSB 2.0

Non-linearity<1%3)

Note 6) The ESD immunity level of contact charge is6K volts, and its air charge is below 8K volts, which is
ESD IEEE std C62.38-1994 compliance.

Operational voltage+12 V DC4)

Power consumption

10 W max

IP classification
Operational temperature

IP67

Note 7) The detector output can adapt for the variation of conveyor speed by using constant integration
time mode with external line trigger.

5)

0 - 40 °C

Relative humidity

30-80 %

Storage temperature-10-50 °C

ESD immunity

6K Volts

On-board calibration

Yes

Constant integration timeYes

Binning function

Yes

AD Gain / Offset calibration

Yes

Averaging and summing function

Yes

Temperature drift compensation

Yes

USB upgrading function

Yes

Support multi USB detectors

Yes

6)

7)

Detection Technology, Inc. | Piisilta 1, 91100 Ii, FINLAND | Tel. +358 8 553 6600 | Fax +358 8 553 6611 | http://www.deetee.com | contact@deetee.com