PC0E-0920

Rugged, stylish body to meet any field requirements!

Handy Type Gas Leak Detector
Model SP-220 TYPE FUM
Model SP-220 TYPE SC
CE Marking-compliant productFor fumigation gas(TYPE FUM)

For semiconductormaterial gas(TYPE SC) Compact and lightweight, with rugged body

Greater shock resistance, protection class equivalent to IP55, with protective cover to preventdirt.

 Quick and reliable detection of low-concentration gases

Newly developed sensor allows detection of low-concentration gases (from 0.1 ppm).

 Button operation to directly detect a wide range of gas concentrations
Switch measurement gases between seven types of fumigation gases and 40 types ofsemiconductor material gases.

 Equipped with data logger function

Up to 256 logs including the date, time and gas concentration of detected gases can be stored.

 With LED light

Take accurate measurements by identifying measurement spots even in dark places.

Specifications
Model
Detection target gas

SP-220 (TYPE FUM)

SP-220 (TYPE SC)

Fumigation gas
See "Detection Target Gas List"on the right

Semiconductor material gas
See "Detection Target Gas List"on the right● SP-220 (TYPE FUM) Detection Target Gas ListGasNo.

Gas Name(Standard Name)

Display

Concentration
Display 1(ppm)Concentration
Display 2(ppm)Concentration
Display 3(ppm)

1Phosphine

PH3

0.1

2.0

4.5

Hot-wire semiconductor

2

Methyl bromide

CH3Br120100Detection range

Depends on detection target gas

3

Carbon disulfide*

CS20.1--

Sampling method

Pump suction type4

Methyl iodide

CH3I

1

1030

Alarm setpoint

Depends on detection target gas

5Hydrogen cyanide HCN

1--6Sulfuryl fluoride--800

7Ethylene bromide* C2H4Br211030Concentration
Display 1(ppm)Concentration
Display 2(ppm)

Detection principle

SO2F2Gas alarm operation

Lamp blinking, intermittent buzzer sounding

Gas alarm reset
operation

Auto-reset

Fault alarm

Sensor abnormality, low battery voltage, low flow rate, system abnormality,
calibration abnormality, clock abnormality, pump abnormality● SP-220 (TYPE SC) Detection Target Gas List

Fault alarm operation

Lamp blinking, intermittent buzzer sounding and fault detail display

Fault alarm reset
operationGasNo.

Self-latching

Response time

Within ten seconds of 0.1 ppm alarm by contact with 0.3 ppm calibration gas (PH₃)

LCD display contents

Operation condition, flow verification, alarm sound, operation sound, battery
level, gas concentration (bar meter), mode, gas name/message

Power supply

AA alkaline battery: 2

Continuous
operating time

Approx. 13 hours (25°C, without alarms or lighting)Operatingtemperature/humidity
range-20 to +55ºC, 95%RH or less (no condensation)

External dimensions

Approx. 43 (W) x 200 (H) x 39 (D) mm (tapered nozzle excluded)

WeightApprox. 215 g (without batteries)

12

Protection class

Equivalent to IP55

1314

Display items• LCD manual backlight (automatically lights up in response to an alarm)• Log data • Peak • Clock

15

Accessories

Hand strap, tapered nozzle, unit protective cover, waterproofing filter forreplacement (5 pcs), 2 AA alkaline dry batteries

Optional(sold separately)

Data logger management program (model: SW-SP-220)* Gases that are prohibited from use in Japan.

* This product is used for detecting ultra-low gas leaks, so the gas concentration values are approximations.

Screen Display
(1)

(7)

(2)

(8)

(3)

(9)
(4)
(10)

(5)
(6)

(1) Operation condition(2) Flow verification(3) Gas concentration 2(4) Gas concentration 1(5) Alarm sound(6) Gas name/message(7) Operation sound(8) Battery level(9) Alarm point(10) Bar meter

Gas Name(Standard Name)

Display1

Phosphine

PH30.1

2.02

Acetone

C3H6O

1103

Arsine

AsH30.2-4

Ammonia

NH310-5

Isobutane

i-C4H101106

Isopropyl alcohol

IPA1107

Carbon monoxide

CO10308Ethyl alcohol

C2H5OH

1109Ethylene

C2H411010Vinyl chloride

VCM

1-11Methyl chloride

CH3CL

110Xylene

C8H10110Ethylene oxide

EO

110Silane

SiH40.5-Methyl bromide

CH3Br12016Hydrogen

H2

11017

Trichloroethylene

C2HCL310-

18Toluene

C7H811019

1,2-Dichloroethane

EDC

11020

Sulfur dioxide

SO21-21Propane

C3H8520

22R-134a

R-134a

50

250

23

R-22

R-22

1050

24

R-32

R-32

105025

Normal hexanen-C6H141050

26Benzene

C6H60.51027Formaldehyde(HCHO)

HCHO

105028Methane

CH412029

Methyl alcohol

CH3OH

11030

Methyl ethyl ketone

MEK11031

Hydrogen sulfide

H2S0.1-32

Diborane

B2H60.1-

33

Germane

GeH40.2-

34

Hydrogen bromide

HBr

10-

35

Hydrogen chloride

HCL

10-

36R-407C

R-407C105037

Hydrogen selenide

H2Se

0.5-

38

R-410AR-410A

1050

39R-404A

R-404A

105040

HFO-1234yf

CH2C2F41030* "-" in the table indicates items that only have a bar display, and do not have ascale.

★Distributed by:

PC0E-0920-220200