SDS2000X Plus
Digital Oscilloscope

Data Sheet

DataSheet-2024.06

EN01D

Key Features
SDS2354X Plus◼

SDS2204X Plus

350 MHz, 200 MHz, 100 MHz models with real-time sample
rate up to 2 GSa/s. A 500 MHz bandwidth upgrade option isavailable for 350 MHz models.

SDS2104X Plus◼

SPO technology


SDS2102X Plus

Waveform capture rates up to 120,000 wfm/s (normalmode) and 500,000 wfm/s (sequence mode)Supports

256-level

intensity

grading

and

color

temperature display modes

Record length up to 200 Mpts/ch, 400 Mpts in total for all
4 channels

Product Overview
SIGLENT’s

SDS2000X

Plus◼
seriesDigitalIntelligent trigger: Edge, Slope, Pulse, Window, Runt,
Interval, Dropout, Pattern and Video (HDTV supported).

Storage

Trigger zone helps to simplify advanced triggering

Oscilloscopes are available in bandwidths of 350 MHz, 200
MHz and 100 MHz, have a maximum sample rate of 2 GSa/s,

Digital trigger system◼

Serial bus triggering and decoder, supports I2C, SPI, UART,

maximum record length of 200 Mpts/ch, and up to 4 analog

CAN, LIN (standard) and CAN FD, FlexRay, I2S, MIL-STD-

channels + 16 digital channels mixed signal analysis ability.

1553B, SENT and Manchester (optional) protocols

The SDS2000X Plus series employs SIGLENT’s SPO◼

Low background noise, features 0.5 mV/div to 10 V/div
voltage scales

technology with a maximum waveform capture rate of up to120,000 wfm/s (normal mode, up to 500,000 wfm/s in◼

10-bit mode provides higher resolution and lower noise

Sequence mode), 256-level intensity grading display function◼

Segmented acquisition (Sequence) mode, dividing theplus a color temperature display mode. It also employs anmaximum record length into multiple segments (up to

innovative digital trigger system with high sensitivity and low

90,000), according to trigger conditions set by the user, withjitter. The trigger system supports multiple powerful triggeringa very small dead time between segments to capture themodes including serial bus triggering. History waveform

qualifying event

recording, Sequence acquisition, Search and Navigate◼

triggered waveforms (frames)

functions allow for extended waveform records to be captured,stored, and analyzed. An impressive array of measurement

History waveform record (History) function for up to 90,000◼

Automatic measurements on 50+ parameters, supports

and math capabilities, options for a 50 MHz waveform

statistics with histogram, track, trend, Gating measurement,

generator, as well as serial decoding, mask test, bode plot, andand measurements on Math, History and Ref

power analysis are also features of the SDS2000X Plus. A 10-◼

average, ERES, and formula editor

bit acquisition mode helps to satisfy applications which require
more than 8-bit resolution.

Two Math traces, support 2 Mpts FFT, +, -, x, ÷, d/dt, ∫dt, √,◼

Abundant data processing and analysis functions such as

The large 10.1’’ capacitive touch screen supports multi-touch

Search, Navigate, Mask Test, Bode plot, Power Analysis

gestures, while the remote web control, mouse and external(optional) and Counter

keyboard support greatly improve the operating efficiency of◼

16 digital channels

the SDS2000X Plus.

◼

Built-in 50 MHz DDS waveform generator (optional)◼

Large 10.1” TFT-LCD display with 1024x600 resolution;
Capacitive touch screen supports multi-touch gestures◼

Multiple interfaces: USB Host, USB Device (USBTMC), LAN（VXI-11/Telnet/Socket）, Pass/Fail, Trigger Out◼

Built-in web server supports remote control by the LAN port
using a web browser; Supports SCPI remote control
commands

WWW.SIGLENT.COM

1

SDS2000X Plus Series Digital Oscilloscope

Models and Key Specifications
Model

SDS2354X Plus

Analog channels

4 + EXT
350 MHz,(upgradable to 500 MHz)

Bandwidth

SDS2104X Plus
SDS2102X Plus
2/4 + EXT

200 MHz

100 MHz

Sample rate (Max.)

2 GSa/s (interleaving mode)，1 GSa/s (non-interleaving mode)

Memory depth (Max.)

200 Mpts/ch (interleaving mode)， 100 Mpts/ch (non-interleaving mode)

Waveform capture rate (Max.)
Vertical resolution
Trigger type
Serial trigger and decode
Measurement
MathData processing and analysis
tools
Digital channel
Waveform generator (optional)Interface
Probe (standard)
Display

2

SDS2204X Plus

Normal mode：120,000 wfm/s;
Sequence mode：500,000 wfm/s
8-bit. 10-bit mode (with typical 100 MHz bandwidth)
Edge, Slope, Pulse, Window, Runt, Interval, Dropout, Pattern, Video and Serial
Standard: I2C, SPI, UART, CAN, LIN
Optional: CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT, Manchester (decode only)More than 50 parameters，supports statistics with histogram, track and trend
2 traces
2 Mpts FFT, +, -, x, ÷, d/dt, ∫dt, √, Identity, Negation, |x|, Sign, ex, 10x, ln, lg, Interpolation,
average, ERES, and formula editor
Search, Navigate, History, Mask test, Bode plot, Power Analysis (optional) and Counter
16-channel; maximum sample rate up to 500 MSa/s; record length up to 50 Mpts/ch
Single channel, frequency up to 50 MHz, 125 MSa/s sample rate, 16 kpts waveform memory
USB 2.0 Host x2, USB 2.0 Device, 10M/100M LAN, External trigger, Auxiliary output (TRIG
OUT，PASS/FAIL)
SP2035A, 350 MHz, 1 probe
PP215, 200 MHz, 1 probe supplied for each channel
supplied for each channel
10.1” TFT-LCD with capacitive touch screen (1024x600)

WWW.SIGLENT.COM

SDS2000X Plus Series Digital Oscilloscope

Functions & Characteristics
Excellent Operability10.1” display with 1024x600 resolution

Capacitive touch screen, supports multi-touch gestures, traces can be moved or scaled efficiently by a finger touch

Built-in web server supports remote control over the LAN port using a web browser

External mouse and keyboard support

Competitive Front End Performance

500 MHz bandwidth (at 2 GSa/s sample rate with 500 MHz

Low noise floor: Only 80 µV rms at 500 MHz bandwidth.

bandwidth option).

WWW.SIGLENT.COM

3

SDS2000X Plus Series Digital Oscilloscope

Up to 120,000 wfm/s waveform update rate

Record Length of up to 200 Mpts/ch

With a waveform update rate of up to 120,000 wfm/s, theUtilizing a hardware-based Zoom technique and record length

oscilloscope can easily capture low-probability events. In

up to 200 Mpts, users can select a slower timebase without

Sequence mode the waveform capture rate can reach 500,000

compromising the sample rate and then quickly zoom in to focus

wfm/s.

on the area of interest.

10-bit Mode

8-bit

10-bit10-bit mode combined with Zoom shows you more details and less noise on the waveform.

4

WWW.SIGLENT.COM

SDS2000X Plus Series Digital Oscilloscope

Multiple Trigger Functions

Trigger Zone

Edge, Slope, Pulse, Video, Windows, Runt, Interval, Dropout,

Trigger Zone is available for advanced triggering.

Pattern and serial trigger.

Measurements for All relevant Parameters and Parameter Statistics

Parameter measurements includes 4 categories: Vertical,

Measurements can be performed within a specified gate period.

Horizontal, Miscellaneous and Channel Delay providing a total

Measurements on Math, Reference and History frames areof 50+ different types of measurements.

supported.
Statistics shows the current value, maximum value, minimum
value, standard deviation and mean value of up to 12
parameters simultaneously. Histogram is available to show the
probability distribution of a parameter. Trend and Track areavailable to show the parameter value vs. time.
In addition, horizontal measurements can process up to 1000
signal edges within one single frame, thus greatly improving thetest efficiency.

WWW.SIGLENT.COM

5

SDS2000X Plus Series Digital Oscilloscope

Advanced Math Function

Two Math traces, support FFT, +, -, x, ÷, d/dt, ∫dt, √, Identity,

Hardware accelerated FFT up to 2 Mpts. This provides high

Negation, |x|, Sign, ex, 10x, ln, lg, Interpolation, average, ERES,

frequency resolution with fast refresh rate. The FFT function

and formula editor.

also supports a variety of window functions so that it can adapt
to different spectrum measurement needs. Three modes(Normal, Average and Max hold) can satisfy different
requirements for observing the power spectrum. Auto peakdetection and markers are supported.

History Mode

Sequence Mode

History function records up to 90,000 triggered waveforms

Segmented memory collection will store the waveform into(frames). This is done continuously in the background, so themultiple memory segments (up to 90,000) and each segment

history waveforms can be played back at any time to find andwill store a triggered waveform together with the dead time

analyze past events. Serial decode, zoom and cursors

information. The interval between segments can be as small as

measurements can be used.

2 µs. All segments can be played back at an arbitrary frame rateusing the History function.

6

WWW.SIGLENT.COM

SDS2000X Plus Series Digital Oscilloscope

Search and Navigate

Serial Bus Decode

The SDS2000X Plus can find events within a record and history

In addition to the decoder lanes correlated to the waveform, bus

based on user specified trigger conditions. Navigate browses

protocol information can be displayed in tabular form. I 2C, SPI,

through Events flagged by the Search, plays back history

UART, CAN, LIN, CAN FD, FlexRay, I2S, MIL-STD-1553B,

frames or continuously moves the delay position on long

SENT and Manchester are supported.

records (useful in zoom view).

Hardware-based High Speed Mask Test Function

The SDS2000X Plus utilizes a hardware-based Mask Test

Built-in Mask Editor application helps to create custom masks.

function, performing up to 80,000 Pass / Fail decisions eachsecond. It is easy to generate user defined test templates which
the signal trace can be continuously compared to. The failed
frames can be automatically stored as history frames or screen
shots, making it suitable for long-term signal monitoring or
automated production line testing.

WWW.SIGLENT.COM

7

SDS2000X Plus Series Digital Oscilloscope

Bode PlotPower Analysis (Optional)

The SDS2000X Plus can control the built-in waveform

The Power Analysis option provides a full suite of power

generator or any stand-alone SIGLENT SDG device to scan themeasurements and analysis, thus improving the efficiency of

amplitude and phase response over frequency of passive or

measurement in switching power supplies and power device

active circuits. The data is presented as Bode Plot. This makes

designs.

it possible to replace expensive network analyzers in lessdemanding applications.

Digital Channels / MSO

50 MHz Built-in Waveform Generator (Optional)

Four analog channels plus 16 digital channels allow theThe built-in DDS waveform generator can output waveforms

acquisition and triggering of mixed waveforms with one

with up to 50 MHz frequency and ±3 V amplitude. Six basic

instrument.

waveforms

togetherwithmultiple

types

of

predefined

waveforms and as user defined arbitrary waveforms aresupported.

8

WWW.SIGLENT.COM

SDS2000X Plus Series Digital Oscilloscope

Complete Connectivity
2 x USB 2.0 Host, 1 x USB 2.0 Device (USBTMC)
1 x 10M/10M LAN (VXI-11/Telnet/Socket)
1 x Auxiliary Output (Pass/Fail and Trigger Out)

WWW.SIGLENT.COM

9

SDS2000X Plus Series Digital Oscilloscope

Specifications
All specifications are not guaranteed unless the following conditions are met:
• The oscilloscope calibration period has not expired• The oscilloscope has been working continuously for at least 30 minutes at the specified temperature (18 ℃ ~ 28 ℃ )Acquisition System (analog channels)
Sample rate

2 GSa/s (interleaving mode* 1)，1 GSa/s (non-interleaving mode* 2 )Memory depth

200 Mpts/ch (interleaving mode) * 3， 100 Mpts/ch (non-interleaving mode) * 3
Normal mode:
120,000 wfm/s max.
Sequence mode: 500,000 wfm/s max.
256 grades
1 ns minimum detectable pulse
90,000 frames max.; Interval between triggers = 2 μs min.
90,000 frames max.
Sin(x)/x, x

Waveform capture rate
Trace intensity
Peak detect
Sequence
History
Interpolation* 1：Interleaving mode: only one of CH1/CH2 and/or only one of CH3/CH4 activated* 2：Non-interleaving mode: both CH1/CH2 and/or both CH3/CH4 activated* 3：In 10-bit mode the maximum memory depth reduces by half

Vertical System

SDS2354X Plus

Analog channels
Bandwidth (-3dB)@ 50 Ω
Rise time (typical)@ 50 Ω
Resolution
Vertical range
Vertical scale(probe 1X)
DC gain accuracy
Offset accuracy

4 + EXT
350 MHz (standard) * 2
200 MHz * 2
500 MHz (optional)* 1,2
1 ns (standard) * 2
1.7 ns * 2
800 ps (optional)* 1,2
8-bit. 10-bit mode (with typical 100 MHz bandwidth)
8 divisions
1 MΩ: 500 μV/div – 10 V/div
50 Ω: 500 μV/div – 1 V/div≤ 3.0%±(1.5%*offset+1.5%*full scale+1 mV)
500 μV/div ~ 100 mV/div: ± 2 V
102 mV/div ~ 1 V/div: ± 20 V
1.02 V/div ~ 10 V/div: ± 200 V
10 kHz ~ BW/10: ±0.5 dB
BW/10 ~ BW/3: ±0.8 dB
BW/3 ~ BW2/3: +1.0 dB, -1.2 dB
BW2/3 ~ BW: +2.0 dB, -2.5 dB
20 MHz (-0, +20%)
200 MHz (-0, +20%)

Offset range (probe 1X)
Bandwidth flatness@ 50 Ω
Bandwidth limit
Low frequency response(AC coupling -3 dB)
Overshoot (150ps fastedge input @50Ω)Coupling
Impedance
Max. Input voltage
SFDR
CH to CH Isolation@ 50Ω
Probe Attenuation

SDS2204X Plus

SDS2104X Plus
SDS2102X Plus
2/4 + EXT
100 MHz3.5 ns * 2

5 Hz (typical)<12% (typical)
DC, AC, GND(1 MΩ ±2%) || (17 pF ±2 pF)
50 Ω: 50 Ω ±1%
1 MΩ ≤400 Vpk(DC + AC), DC~10 kHz
50 Ω ≤5 Vrms, ±10 V Peak
≥40 dB
DC～100 MHz: >40 dB
100 MHz ~ BW: ≥34 dB
1X, 10X, 100X, Custom* 1：In interleaving mode bandwidth is 500 MHz, rise time is 0.8 ns; in non-interleaving mode bandwidth is 350 MHz, rise time is 1 ns
* 2：In 10-bit mode bandwidth is 100 MHz (typical)，rise time is 3.3 ns (typical)

10

WWW.SIGLENT.COM

SDS2000X Plus Series Digital Oscilloscope

Horizontal System
Time scale
Horizontal range
Display mode
Skew (CH1~CH4)
Time base Accuracy

1 ns/div – 1000 s/div
0.5 ns/div – 1000 s/div when 500 MHz bandwidth option is installed
10 divisions
Y-T, X-Y, Roll (≥50 ms/div)<100 ps±1ppm initial; ±1ppm 1st year aging; ±3.5ppm 10-year aging

Trigger System
Mode
Level
Ext Trigger Channel
input voltage
Hold off range

Coupling

Accuracy (typical)

Auto, Normal, Single
Internal: ±4.1 div from the center of the screen
EXT: ±0.61 V
EXT/5: ±3.05 V
1 MΩ ≤ 42 Vpk
50 Ω ≤ 5 Vrms
By time：8 ns ～ 30 s (8 ns step)
By event：1 ~ 108
CH1~CH4
DC: Passes all components of the signal
AC: Blocks DC components and attenuates signals below 20 Hz
LFRJ: Attenuates the frequency components below 1.2 MHz
HFRJ: Attenuates the frequency components above 600 kHz
Noise RJ: Increases the trigger hysteresis
EXT
DC: Passes all components of the signal
AC: Blocks DC components and attenuates signals below 8 Hz
LFRJ: Attenuates the frequency components below 33 kHz
HFRJ: Attenuates the frequency components above 967 kHz
CH1～CH4: ±0.2 div
EXT: ±0.3 div
CH1 ～ CH4:

Sensitivity

EXT:

EXT/5:
Jitter
Displacement
Zone
Edge Trigger
SourceSlope
Slope Trigger
SourceSlope
Limit range>10 mV/div:
5 mV/div~10 mV/div:≤ 2 mV/div:

Noise RJ = OFF±0.26 div
±0.26 div±0.5 div

Noise RJ = ON
±0.33 div
±0.33 div±0.5 div

200 mVpp, DC ～ 10 MHz
300 mVpp, 10 MHz ～ 300 MHz
1 Vpp, DC ～ 10 MHz
1.5 Vpp, 10 MHz ～ 300 MHz

CH1～CH4: <10 ps rms, 6 divisions pk-pk, 2 ns edge
EXT: <200 ps rms
Pre-Trigger: 0 ~ 100% memory
Delay-Trigger: 0 ~ 5,000 div
Up to 2 zones; Source: CH1~CH4; Property: Intersect, Not Intersect
CH1~CH4/EXT/(EXT/5)/AC Line/D0~D15
Rising, Falling, Rising & Falling
CH1~CH4
Rising, Falling≤, ≥, in range, out of range

Time range

2 ns ～ 20 s，1 ns resolution

Pulse Width Trigger
SourcePolarity
Limit range

CH1~CH4/D0~D15+wid, -wid≤, ≥, in range, out of range

Time range

2 ns ～ 20 s，1 ns resolution

Video Trigger
SourceCH1~CH4

Standard

NTSC，PAL，720p/50，720p/60，1080p/50，1080p/60，1080i/50，1080i/60，Custom

WWW.SIGLENT.COM

11

SDS2000X Plus Series Digital Oscilloscope
Synchronization
Trigger Condition
Window Trigger
SourceWindow typeInterval Trigger
SourceSlope
Limit range

Any, SelectLine, Field

Time range

2 ns ～ 20 s，1 ns resolution

Dropout Trigger
SourceTimeout type
Slope

CH1~CH4/D0~D15
Edge, State
Rising, Falling

Time range

2 ns ～ 20 s，1 ns resolution

Runt Trigger
SourcePolarity
Limit range

CH1~CH4
Positive, Negative≤, ≥, in range, out of range

Time range

2 ns ～ 20 s，1 ns resolution

Pattern Trigger
SourcePattern Setting
Logic
Limit range

CH1~CH4/D0~D15
Don’t Care, Low, HighAND, OR, NAND, NOR≤, ≥, in range, out of range

Time range

2 ns ～ 20 s，1 ns resolution

Serial Trigger
SourceCH1~CH4/D0~D15

Protocol

CH1~CH4
Absolute, Relative
CH1~CH4/D0~D15
Rising, Falling≤, ≥, in range, out of range

Standard: I2C、SPI、UART、CAN、LIN
Optional: CAN FD、FlexRay、I2S、MIL-STD-1553B

I2C trigger
SPI trigger
UART trigger
CAN trigger
LIN trigger
CAN FD trigger(optional)
FlexRay trigger(optional)
I2S trigger (optional)
MIL-STD-1553B trigger(optional)

Type: Start, Stop, Restart, No Ack, EEPROM, Address & Data, Data Length
Type: Data
Type: Start, Stop, Data, Parity Error
Type: All, Remote, ID, ID+Data, Error
Type: Break, Frame ID, ID+Data, Error

SENT (Optional)

Type：Start, Slow channel, Fast channel, Error

Type: Start, Remote, ID, ID+Data, Error
Type: TSS, Frame, Symbol, Errors
Type: Data, Mute, Clip, Glitch, Rising Edge, Falling Edge
Type: Transfer, Word, Error, Timing

Serial Decoder
Decoders
Decoder Type

2
Full duplex

Threshold-4.1 ～ 4.1 div

List

1 ～ 7 lines

I2 C
SignalSCL，SDA

Address

7bit，10bit
2,000

Decoded frames (Max.)
SPI
SignalEdge SelectChip selectBit Order
12

CLK，MISO，MOSI，CS
Rising, Falling
Active high, active low, clock timeout
LSB, MSB
WWW.SIGLENT.COM

SDS2000X Plus Series Digital Oscilloscope
Decoded frames (Max.) 15,000
UART
SignalRX，TX

Data Width

5 bit，6 bit，7 bit，8 bit
None, Odd, Even, Mark, Space

Parity Check
Stop Bit

1 bit，1.5 bit，2 bit
Idle Level
Low, high
Bit Order
LSB, MSB
Decoded frames (Max.) 15,000
CAN
SourceCH1~CH4/D0~D15
Decoded frames (Max.) 2,000
LIN
LIN
Specification
Ver1.3，Ver2.0
Package Revision
Baud Rate
600 bps, 1200 bps, 2400 bps, 4800 bps, 9600 bps, 19200 bps, custom
Decoded frames (Max.) 3,000
CAN FD (optional)
SourceCH1~CH4/D0~D15
Nominal Baud Rate
10 kbps, 25 kbps, 50 kbps, 100 kbps, 250 kbps, 1 Mbps, custom
Data Baud Rate
500 kbps, 1 Mbps, 2 Mbps, 5 Mbps, 8 Mbps, 10 Mbps, custom
Decoded frames (Max.) 1,000
FlexRay (optional)
SourceCH1~CH4
Data Baud Rate
2.5 Mbps, 5 Mbps, 10 Mbps, custom
Decoded frames (Max.) 1,000
I2S (optional)
Signal
BCLK, WS, DATA
Audio Variant
Audio-I2S, Audio-LJ, Audio-RJ
Start Bits
0~31
Baud Rate
1~32
Decoded frames (Max.) 10,000
MIL-STD-1553B (optional)
SourceCH1~CH4
Decoded frames (Max.) 10,000
SENT (Optional)
SourceCH1~CH4/D0~D15
Manchester (Optional)
SourceCH1~CH4
Baud Rate
500 bps~5 Mbps

Measurement
Auto measurement
SourceMode
Range
VerticalHorizontal
Miscellaneous
Delay
Statistics
Cursors
SourceType

WWW.SIGLENT.COM

CH1~CH4, D0~D15, F1~F2, Ref, History, Z1~Z4
Simple, Advanced
Screen, Gate
Max, Min, Pk-Pk, Top, Base, Amplitude, Mean, Cycle Mean, Stdev, Cycle Stdev, RMS, Cycle RMS,Median, Cycle Median, FOV, FPRE, ROV, RPRE, Level@Trigger
Period, Frequency, Time@max, Time@min, +Width, -Width, 10-90%Rise time, 90-10%Fall time, Risetime, Fall time, +Burst Width, -Burst Width, +Duty Cycle, -Duty Cycle, Delay, Time@Middle, Cycle-Cycle
jitter+Area@DC, -Area@DC, Area@DC, Absolute Area@DC, +Area@AC, -Area@AC, Area@AC, Absolute
Area@AC, Cycles, Rising Edges, Falling Edges, Edges, Positive pulses, Negative pulses, Positive
Slope, Negative Slope
Phase, FRFR, FRFF, FFFR, FFFF, FRLR, FRLF, FFLR, FFLF, Skew, tsu@R, tsu@F, th@R, th@F
Current, Mean, Min, Max, Sdev, Count, Histogram, Trend, Track
CH1~CH4, D0~D15, Math, Ref
Manual: Time X1, X2, (X1-X2), (1/ΔT)
Voltage/Current: Y1, Y2, (Y1-Y2)
Track: Time X1, X2, (X1-X2)
Measure: indicates the measurement on specific parameter13

SDS2000X Plus Series Digital Oscilloscope

Math
Traces
SourceOperation

FFT

F1, F2
CH1~CH4, Z1~Z4, F1~F2+, -, *, ÷, FFT, d/dt, ∫dt, √, Identity, Negation, |x|, Sign, ex, 10x, ln, lg, Interpolation, Average, ERES,
Formula Editor
Length: 2 Mpts, 1 Mpts, 512 kpts, 256 kpts, 128 kpts, 64 kpts, 32 kpts, 16 kpts, 8 kpts, 4 kpts, 2 kpts
Window: Rectangular, Blackman, Hanning, Hamming, Flattop
Display: Full Screen, Split, Exclusive
Mode: Normal, Max hold, Average
Tools: Peaks, Markers

Analysis
SearchSourceMode
Copy setting
Navigate
Type
Mask Test

CH1~CH4, History
Edge, Slope, Pulse, Interval, Runt
Copy from trigger, Copy to trigger
Search event, Time, History frame

SourceCH1~CH4，Z1~Z4

Mask creating
Mask test speed
Store failed frames
Bode Plot
SourceSupported signal
sources
Sweep type

Auto (Create mask), Custom (Mask Editor)
Up to 80,000 frames/s
To history, To screenshot

CH1~CH4
Built-in waveform generator
SDG series waveform generators, Connection: USB, LAN
Simple, Vari-level
Mode: Linear, Logarithmic
Frequency
Range: 10 Hz ~ 120 MHz
MeasureUpper cutoff frequency, Lower cutoff frequency, Bandwidth, Gain margin, Phase margin
Power Analysis (Optional)
Power quality, Current Harmonics, Inrush current, Switching loss, Slew rate, Modulation, Output ripple,
MeasureTurn on/turn off, Transient response, PSRR, Efficiency
Counter
SourceCH1~CH4
Frequency resolution
7 digits
Totalizer
Counter on edges, support Gate and Trigger

500 MHz Bandwidth Extension (optional)
Channels
Bandwidth (-3dB)@50 Ω
Rise time (typical)@50 Ω
Sample Rate
Resolution
Memory Depth

2 (CH1&CH3, CH1&CH4, CH2&CH3 or CH2&CH4)
500 MHz800 ps
2 GSa/s
8-bit. 10-bit mode (with typical 100 MHz bandwidth)
200 Mpts/ch

Digital Channels
ChannelsMax. Sampling Rate
Memory Depth
Min. Detectable Pulse
Level Range
Logic TypeSkew

16, divided to 2 groups: D0~D7, D8~D15
500 MSa/s
50 Mpts/ch
3.3 ns-10 V~10 V
TTL, CMOS, LVCMOS3.3, LVCMOS2.5, Custom
D0~D15: ±1 sampling interval
Digital to Analog: ± (1 sampling interval +1 ns)

Waveform Generator (optional)
ChannelsMax. Output Frequency
Sampling Rate
Frequency Resolution
14

1
50 MHz
125 MSa/s
1 μHz
WWW.SIGLENT.COM

SDS2000X Plus Series Digital Oscilloscope
Frequency Accuracy±50 ppm
Vertical Resolution
14 bit-1.5 V ~ +1.5 V (into 50Ω)Amplitude Range-3 V ~ +3 V (into High-Z)
Waveforms
Sine, Square, Ramp, Pulse, DC, Noise, 45 Arbitrary
Output Impedance
50 Ω ±2%
Protection
Over voltage protection, Current limit
Sine
Frequency
Offset accuracy
kHz)

1 μHz ～ 50 MHz(10

Amplitude flatness

SFDR

Harmonic distortion±(1%*offset setting value +3 mVpp)
Compare to 10 kHz, 2.5 Vpp (into 50Ω):±0.3 dB，≤25 MHz±0.5 dB，>25 MHz
DC~1 MHz: -60 dBc
1 MHz~5 MHz: -55 dBc
5 MHz~25 MHz: -50 dBc
25 MHz~50 MHz: -40 dBc
DC~5 MHz: -50dBc
5 MHz~25MHz: -45dBc
25 MHz~50 MHz: -40 dBc

Square/Pulse
Frequency

1 μHz ～ 10 MHz

Duty cycle

1% ～ 99%

Edge< 24 ns (10% ～ 90%)

Overshoot< 3% (typical，1 kHz, 1 Vpp)
> 50 ns< 500 ps + 10 ppm

Pulse width
Jitter (cycle-cycle)
Ramp
Linearity

1 μHz ～ 300 kHz< 0.1% of Pk-Pk (typical, 1 kHz, 1 Vpp, 50% symmetry)

Channels

0% ～ 100%

Frequency

DC
Offset range
Accuracy
Noise
Bandwidth (-3 dB)
Arb
Frequency
Waveform memory
Sample rate
Wave import±1.5 V (into 50 Ω)±3 V (into Hi-Z)±(|setting value|*1%+3 mV)>25 MHz
1 μHz ～ 5 MHz
16 kpts DDS memory
125 MSa/s
From EasyWaveX, from U-disk, directly from waveform data of analog channels

I/O
Front panel

Rear panel

USB 2.0 Host x2
Probe compensation: 1 kHz, 3 VPP square wave
USB 2.0 Device
LAN: 10M/100M
EXT trigger: EXT ≤1.5 Vrms，EXT/5 ≤7.5Vrms
Auxiliary output: TRIG OUT -- 3.3 V LVCMOS; PASS/FAIL OUT -- 3.3 V TTL

Display
Display Type
Resolution
Contrast
Backlight

10.1”TFT LCD with capacitive touch screen
1024×600
500:1 typical
500 nit typical

Display Setting
Range
WWW.SIGLENT.COM

8 x 10 grid15

Display type
Persistence Time
Color Display
Language
Built-in help

SDS2000X Plus Series Digital Oscilloscope
Dot, Vector
OFF, 1 s, 5 s, 10 s, 30 s, infinite
Normal, Color; Supports customer trace color
Simplified Chinese, Traditional Chinese, English, French, Japanese, German, Spanish, Russian,
Italian, Portuguese
Simplified Chinese, English

Environmental
Temperature
Humidity
AltitudeElectromagnetic
Compatibility

SafetyOperating: 0 °C ~ 40 °C
Non-operating: -20 °C ~ 60 °C
Operating: 85% RH, 40 °C , 24 hours
Non-operating: 85% RH, 65 °C, 24 hours
Operating: ≤3,000 m
Non-operating: ≤15,000 m
Meets EMC directive (2014/30/EU), meets or exceeds IEC 61326-1:2012/EN61326-1:2013 (Basic)
Conducted disturbance

CISPR 11/EN 55011

CLASS A group 1 ， 150kHz30MHz

Radiated disturbance

CISPR 11/EN 55011

CLASS A group 1 ， 30MHz1GHz

Electrostatic discharge (ESD)

IEC 61000-4-2/EN 61000-4-2

Radio-frequency
electromagnetic field Immunity

IEC 61000-4-3/EN 61000-4-3

Electrical fast transients (EFT)IEC 61000-4-4/EN 61000-4-4

SurgesIEC 61000-4-5/EN 61000-4-5

Radio-frequency
continuous
conducted Immunity

IEC 61000-4-6/EN 61000-4-6

4.0 kV (Contact)，8.0 kV (Air)
10 V/m (80 MHz to 1 GHz);
3 V/m (1.4 GHz to 2 GHz);
1 V/m (2.0 GHz to 2.7GHz)
2kV (Input AC Power Ports)
1kV (Line to line)
2kV (Line to ground)
3 V, 0.15-80MHz

Voltage Dips：
0% UT during half cycle;
0% UT during 1 cycle;
Voltage dips and interruptions
IEC 61000-4-11/EN 61000-4-11
70% UT during 25/30 cycles
Voltage interruptions ： 0% UTduring 250/300 cycles
UL 61010-1:2012/R: 2018-11; CAN/CSA-C22.2 No. 61010-1:2012/A1:2018-11.
UL 61010-2-030:2018; CAN/CSA-C22.2 No. 61010-2-030:2018.

Power Supply
Input Voltage &Frequency
Power consumption

100 ～ 240 Vrms 50/60Hz
80 W max., 50 W typical, 4 W typical in standby mode

Mechanical
Dimensions

Length x Height x Width = 352 mm×224 mm×111 mm

Weight

Net Weight: 3.3 kg (2-ch); 3.9 kg (4-ch)
Gross Weight: 4.8 kg (2-ch); 5.4 kg (4-ch)

16

WWW.SIGLENT.COM

SDS2000X Plus Series Digital Oscilloscope

Ordering Information
Model

Description

SDS2354X Plus

350 MHz，4-ch，2 GSa/s (Max.)，200 Mpts，10.1”touch screen

SDS2204X Plus

200 MHz，4-ch，2 GSa/s (Max.)，200 Mpts，10.1”touch screen

SDS2104X Plus

100 MHz，4-ch，2 GSa/s (Max.)，200 Mpts，10.1”touch screen

SDS2102X Plus

100 MHz，2-ch，2 GSa/s (Max.)，200 Mpts，10.1”touch screenStandard Accessories

Quantity

USB cable
Quick start
Passive probe
Certificate of calibration
Power cord

11x2 (2-ch model); x4 (4-ch model)
1
1

Optional Accessories

Part Number

16-channel logic probe
Power Analysis deskew fixture
STB3 demo signal source
High voltage probe
High voltage differential probe
Current probeBag

SPL2016
DF2001A
STB3
HPB4010
DPB1300/DPB4080/DPB5150/DPB5150A/DPB5700/DPB5700A
CPL5100/CP4020/CP4050/CP4070/CP4070A/CP6030/CP6030A/CP6150/CP6500
BAG-S2

Options

Part Number

Waveform generator option (software)Power Analysis (software)
I2S trigger & decode (software)
MIL-STD-1553B trigger & decode (software)
FlexRay trigger & decode (software)
CAN FD trigger & decode (software)
SENT trigger & decode (software)
Manchester decode (software)
100 MHz to 200 MHz bandwidth upgrade (4-ch model) (software)
200 MHz to 350 MHz bandwidth upgrade (4-ch model) (software)
350 MHz to 500 MHz bandwidth upgrade (4-ch model) (software)
100 MHz to 350 MHz bandwidth upgrade (2-ch model) (software)

SDS2000XP-FG
SDS2000XP-PA
SDS2000XP-I2S
SDS2000XP-1553B
SDS2000XP-FlexRay
SDS2000XP-CANFDSDS2000XP-SENT
SDS2000XP-Manch
SDS2000XP-4BW02
SDS2000XP-4BW03
SDS2000XP-4BW05
SDS2000XP-2BW03

WWW.SIGLENT.COM

17

SDS2000X Plus Series Digital Oscilloscope

About SIGLENTSIGLENT is an international high-tech company, concentrating on R&D, sales,
production and services of electronic test & measurement instruments.

SIGLENT first began developing digital oscilloscopes independently in 2002.
After more than a decade of continuous development, SIGLENT has extended
its product line to include digital oscilloscopes, isolated handheld

oscilloscopes,

function/arbitrary

waveform

generators,

RF/MWsignalgenerators, spectrum analyzers, vector network analyzers, digital multimeters,
DC power supplies, electronic loads and other general purpose test
instrumentation. Since its first oscilloscope was launched in 2005, SIGLENT
has become the fastest growing manufacturer of digital oscilloscopes. We
firmly believe that today SIGLENT is the best value in electronic test &measurement.

Headquarters:
SIGLENT Technologies Co., Ltd
Add: Bldg No.4 & No.5, Antongda Industrial

Zone, 3rd Liuxian Road, Bao'an District,
Shenzhen, 518101, China
Tel: + 86 755 3688 7876
Fax: + 86 755 3359 1582
Email: sales@siglent.com
Website: int.siglent.com

North America:
SIGLENT Technologies America, Inc
6557 Cochran Rd Solon, Ohio 44139
Tel: 440-398-5800

Toll Free: 877-515-5551
Fax: 440-399-1211
Email: info@siglent.com

Follow us on
Facebook: SiglentTech

Website: www.siglentna.com

Europe:SIGLENT Technologies Germany GmbH
Add: Staetzlinger Str. 70
86165 Augsburg, Germany
Tel: +49(0)-821-666 0 111 0
Fax: +49(0)-821-666 0 111 22

Email: info-eu@siglent.com

18

Website: www.siglenteu.com

WWW.SIGLENT.COM