INST.NO. INE-893B

DB600
Setting Software

Table of contents
1. Introduction .........................................................................................................................1
2. System Requirement ..........................................................................................................3

2-1 Operation Condition of the Software ...................................................................................................... 3

3 How to Setup ........................................................................................................................4

3-1. Installation .............................................................................................................................................. 4
3-1-1. New installation ............................................................................................................................... 4
3-1-2. Installation at version upgrade........................................................................................................ 7
3-2. Uninstallation.......................................................................................................................................... 8

4. Startup and Exit of the Software .....................................................................................10

4-1. Startup .................................................................................................................................................. 10
4-2. Exit .........................................................................................................................................................11

5. How to Operate .................................................................................................................125-1. How to operate ..................................................................................................................................... 12
5-2. Operation of Home............................................................................................................................... 14
5-2-1. Port Registration Window ............................................................................................................. 15
5-2-2. Device Registration Window ........................................................................................................ 17
5-2-3. Summary Display Window............................................................................................................ 25
5-2-4. Version Information Dialog ........................................................................................................... 29
5-3. Operation of Parameter Setting........................................................................................................... 30
5-3-1. Parameter Setting Window ........................................................................................................... 31
5-3-2. Parameter Setting Window (Off-Line) .......................................................................................... 61
5-4. Operation of Acquisition ....................................................................................................................... 69
5-4-1. Data Registration Window ............................................................................................................ 71
5-4-2. Group Registration Window ......................................................................................................... 73
5-4-3. Acquisition Group Management Window ..................................................................................... 75
5-4-4. Alarm Display Window .................................................................................................................. 76
5-4-5. Data Memory Operation Window ................................................................................................. 77
5-4-6. Trend Graph Display Window....................................................................................................... 79
5-4-7. List of Real Time Data Display Window ....................................................................................... 85
5-5. Operation of Data Analysis .................................................................................................................. 87
5-5-1. Data Analysis Window .................................................................................................................. 88
5-6. Operation of Favorite ......................................................................................................................... 108

6. Troubleshooting.............................................................................................................. 112

1. Introduction
Thank you for using our DB600 Setting Software.
This software provides application software function of parameter setup, data acquisition, data analysis toa DB600 device and linking up each device.
This instruction manual describes how to prepare hardware, program installation and operation etc. Make
sure to read this instruction manual in advance in order to understand this software well and to prevent
troubles from occurring.

Note
1.

Scope
The following license terms apply to the CHINO product you are using this time.
2. CopyrightThis software is under the protection of the copyright law, international copyright treaties andother law and treaties related to intangible property right. The intellectual property rights such as
patent, copyright, commercially confidential matter and trademark are belonging to CHINO.
3. Scope of license
The software may be used only for the device you purchased. Within the scope of use, thesoftware may be installed on more than one PC by more than one user.
4. Prohibition of use by the third party
You may not permit the use of this software by licensing, devolving, distributing or renting to thethird party.
5. Restriction on copying
You may only make a copy of this software which is provided in the form of a storage medium
for backup usage.
6. Prohibition of modification
You may not alter or modify a part of this software and accessory (including partial integration ofthis software to other software).
7. Warranty
Since this software is provided for free of charge, CHINO will not support or perform operation
guarantee.
CHINO is not responsible for repair of the software failure.
8. Limitation of liability
CHINO is not responsible for any loss or damages caused directly or indirectly by operation ofthis software and for any conflict arise by this software between you and the third party.
9. Operation on power failureThis software stops when the power is impossible to be supplied, at a time such as power
failure. Also, please note that, it is required to start up manually after the recovery from power
failure.
10. OtherDue to improvement or for some other reasons, the specifications of this software may bealtered by CHINO without prior notice. There may be differences between the image of thismanual and the actual image on the screen.

-1-

Notice
1.
2.
3.

4.

No part of this manual can be reproduced or copied in any form without permission.
The contents of this manual may be altered without prior notice.
This manual has been documented by making assurance doubly sure. However, if any question
arises or if any error, an omission, or other deficiencies are found, please contact your nearest
our sales office.
CHINO is not responsible for any operation results of this software

Trademarks
1. Microsoft, Windows, Windows 7, Windows 8, Windows 10, .NET Framework, and Office aretrademarks of Microsoft Corporation and the related company.
2. Other described company names and product names are trademarks and registered products of
the respective companies.
3. Please note that the marks “TM” and “®” are omitted throughout this manual.

Precautions
1.
2.

Keep this instruction manual carefully until the software is discarded.
When discarding the software, follow the local regulations for waste disposal and cooperate inrecycling.

-2-2. System Requirement
Use the software in the environment described below.

2-1 Operation Condition of the Software
Required
instruments

Contents and condition

PC

CPU
MemoryHard disk space
Removable disk drive
Supported OS

Required library.NET Framework4
Microsoft Office2007/2010*If these are not installed, a part of file output function is to be restricted.
Screen resolution 1024x700 or more.

Display

1GHz 32 bit or 64 bit.
1GB or more (32bit), 2GB or more (64bit).
2GB or more free hard disk space.
Compatible with CF
Windows 7 (32bit/64bit) English
Windows 8 (32bit/64bit) English
Windows 10 (32bit/64bit) English*.NET Framework4 or later need to be able to installed to
the OS above.

-3-

3 How to Setup
3-1. Installation
3-1-1. New installation
Install the software from install medium to a PC before use.
Follow the procedures for installation below.

< Procedures >(1) Start up installer.

Start Windows then start “setup.exe” from install medium.

(2) License agreement.

Installation of “Microsoft .NET Framework4” is required in advance.
Click [Accept] button.
*If it is installed, the dialog may not be
displayed.

*Note on license agreement
If this dialog is displayed, check that the
PC is connected to the network. Executethe installation again in the condition ofconnected to the network.

-4-(3) Start the Installation Wizard.

DB600 Setting Software Setup Wizard
dialog is started. Click [Next] button.

(4) Select the Installation Folder.

On Select Installation Folder dialog, select[the install folder] and [user]. Click [Next]
button.

(5) Confirm the installation.

On Installation Confirmation dialog, click[Next] button.

-5-(6) Start the installation.

The installation is started.
The dialog indicating the progress of theinstallation appears. Wait until the
installation completes.

(7) Complete the installation.

Installation Complete dialog appears.
Click [Close] button to finish.

-6-3-1-2. Installation at version upgradeThis software may upgrade to a new version to add new functions and/or fix failures.
Follow the procedures for version upgrade below.

< Procedures >(1) Uninstall the current version (refer to the section 3-2).
(2) Install a new version (refer to the section 3-1-1).

-7-

3-2. Uninstallation
This section describes how to delete the software from the hard disk.
Before starting the uninstallation, exit from all the programs that are related to this software.

< Procedures >(1) Open a Control Panel.
Click in following order[Start] → [Control Panel].

(2) Click [Uninstall a program].
After the Control Panel appears, click[Uninstall a program].

(3) Delete [DB600 Setting Software].
Select [DB600 Setting Software from a
list, then click [Uninstall].

-8-

(4) Click [Yes].
Click [Yes], on Programs and Featuresdialog.

(5) Start the uninstallation.
The uninstallation is started. The dialog
indicating the progress of theuninstallatSerrion appears. Wait until theuninstallation completes. When theuninstallation is completed, the dialog
shown right is closed automatically.
*At this point, a folder related to thesoftware still remains. To delete thefolder completely, delete the related
folder “DB600 Setting Software” byWindows Explorer. The location of thefolder is shown in the table below【Location of the folder related to the software】
OS
Windows 7
Windows 8
Windows 10

Location of the folder
C:¥ProgramData¥ DB600 Setting Software
C:¥ProgramData¥ DB600 Setting Software
C:¥ProgramData¥ DB600 Setting Software-9-4. Startup and Exit of the Software
4-1. Startup
This section describes how to startup the software.

< Procedures >(1) Display [All Programs].
Click [Start] → [All Programs].

(2) Startup DB600 Setting Software.
Next, click[CHINO] → [DB600 Setting Software]to start up the software.

- 10 -

4-2. Exit
This section describes how to exit the software.

< Procedures >Click [ X ] on the right end of the title bar
and close the Main Operation window.

- 11 -

5. How to OperateThis section describes how to operate the application.
*The images used in the description are in the development phase. Please note that there may be differences
between the actual images on the screen and images on this document.

5-1. How to operate< Procedures >(1) Start this software.

When you startup, application launcher for the main operation window is displayed.

(2) Select a function.

Five main functions are available at the application launcher.
From the left of menu bar, there are [Favorite], [Home], [ParameterSetting], [Acquisition] and [Analysis]. Byclicking the each menu button, tool buttons below change to the relevant functions.
Click [Acquisition] button.

A list of data acquisition function tool
buttons is displayed.

- 12 -

(3) Startup of each function.

After (2) Select a function, click tool button below.
By clicking the tool button, relevant function and/or window/dialog is started up.
*Maximum 10 screen can be opened. However following screen is excluded.
･ “5-2-1 Port Registration Window”･ “5-2-2 Device Registration Window”･ “5-2-4 Version Information Dialog”･ “5-4-1 Data Registration Window”･ “5-4-2 Group Registration Window”
Click port registration button.

Clicked tool button relevant window orfunction is displayed or performed.

(4) Exit this software.

Close the application launcher for main operation window.

Remarks

Precautions at exit

If close the application launcher while setting/acquisition/analysis is running, each of relevant function
and window is also closed. For exit of the application, it is recommended to follow the transition of exit
the relevant function first then exit the application launcher.

- 13 -

5-2. Operation of Home
Home provides whole application related registration of setting information and monitoring of condition etc.

< Name of home parts at application launcher >Utilizing following functions (1) Port Registration, (2) Device Registration, (3) Summary Display,(4) Version Information Display, it manages basic setting information of this application.

(1)

No.

(2)

(3)

(4)

Name

(1)

Port Registration(2)

Device Registration

Description

Reference

Displays port registration window. It provides registration, editing and

5-2-1

delete function of port setting.
Displays device registration window. It provides registration, editing

5-2-2

and delete function of connecting device setting.
(3)

Summary Display

Displays a list of summary. It also provides setting related to display.

5-2-3

(4)

Version Information

Displays a list of version information of this application and DLL of

5-2-4

Displayholding devices.

- 14 -

5-2-1. Port Registration Window
It provides registration, editing and deleting of setting information for serial port communication of theapplication.

< Name of port registration window parts >Utilizing followings (1) New Registration, (2) Edit, (3) Delete, (4) List of Registered Port Display, (5) OK and(6) Cancel, it sets serial port communication.

(2)

(1)

(3)

(4)

(5)

No.
(1)
(2)

Name

Description

New

Displays port editing dialog. Adds new port to list of port information.

Registration*Only one port can be registered.

Edit

From registered port setting information, displays port editing dialog.

(6)

Reference
5-2-1-1
5-2-1-1*Only for the edit, double click (4) List of Registered Port Display enables
displaying port editing dialog.
(3)

DeleteFrom registered port setting information, delete selected port.

-*Deleting multiples is available.
(4)

List of

Displays currently registered port setting information.

-

After starting up port registration window, reflect contents of new-RegisteredPort Display(5)

OK

registration edit and delete.
(6)

CancelAfter starting up port registration window, cancel contents of new
registration edit and delete.

- 15 --

5-2-1-1. Port Edit Dialog

It provides editing function of port registration window; port setting information.

< Name of port edit dialog parts >Utilizing followings (1) Port Number, (2) Transmission Mode, (3) Baud Rate, (4) Data Bit, (5) Parity,(6) Stop Bit, (7) OK and (8) Cancel, it edits or cancels for a port setting information.

(1)

(2)

(3)

(4)

(5)

(6)
(7)

No.

Name

(1)

Port Number(8)
Description

Select a port number from the range of COM1 to COM256.
*Icon shown below is displayed for the available port number.

(2)

Transmission

Select either RTU or ASCII.

Mode
(3)

Baud Rate

Select from 9600, 19200 or 38400.

(4)

Data Bit

Select either 7 or 8.

(5)

ParitySelect from None, Even or Odd.

(6)

Stop Bit

Select either 1 or 2.

(7)

OK

Reflects edited port setting information on the port registration window.

(8)

CancelCancels edited port setting information. If it is new registration, cancels add and if it isediting of registered port setting, returns to the prior to the editing.

- 16 -

5-2-2. Device Registration Window
It provides registration, editing and deleting of communication device setting information in the application.

< Name of device registration window parts >Utilizing followings (1) Model Code Direct Entry, (2) Wizard Registration, (3) Device Scan, (4) RegisteredDevice Detail Setting, (5) Copy, (6) Cut, (7) Paste, (8) Delete, (9) List of Registered Device Display,(10) OK and (11) Cancel, it sets device registration information.

(1) (2) (3) (4) (5) (6) (7) (8)(9)(10) (11)

No.

Name

Description

Reference(1)

Model CodeDisplays model code entry dialog. Enter model code of desired device

5-2-2-1

Direct Entry

directly.

WizardStarts device registration wizard. Specify desired device and each

Registration

settings etc. from selections.

(3)

Device Scan

Displays device scan dialog. Specify communication settings and(4)Registered Device Displays device setting editing dialog. Edit detail setting information of(2)

5-2-2-2
5-2-2-3

obtains model code from currently connecting device.
Detail Setting

device which model code is fixed.

(5)

Copy

Copies device information of select area in the list of registered device.

(6)

Cut

5-2-2-4-*Shortcut is available with Ctrl + C key.
Cuts device information of select area in the list of registered device.

-*Shortcut is available with Ctrl + X key.
(7)

Paste

(8)

DeletePastes holding device information obtained by (5) Copy or (6) Cut-

after the selected row. *Shortcut is available with Ctrl + V key.
From registered device information, deletes selecting device-

information. *Deleting multiples is available.
(9)
(10)

List of RegisteredDisplays registered device setting information at present.

Device Display*Up to 10 devices are able to be registered.

OK

After starting up device registration window, reflects contents of new-

registration edit and delete.
(11)

CancelAfter starting up device registration, cancels contents of new
registration edit and delete.

- 17 --

5-2-2-1. Model Code Entry Dialog

It provides registration function of device information from device registration window; model code directentry.

< Name of model code entry dialog parts >Utilizing followings (1) Model Code Entry, (2) OK and (3) Cancel, it registers or cancel device information.

(1)
(2)

No.
(1)

(3)

Name
Model Code Entry

Description
Enter desired model code directly.

Block
(2)

OK

Creates and register device information from entered model code.
*If entered model code is not included in registerable model code, message of invalid
registration appears.

(3)

CancelIgnoring entered model code, exit model code entry dialog.

- 18 -

5-2-2-2. Device Registration Wizard

It provides function of registration for device information from device registration window; wizard.

< 1. Name of device registration wizard (setting of each device) parts >Utilizing followings (1) Model, (2) Model Selection, (3) Next and (4) Cancel, it progresses, regresses orcancels device registration wizard.

(1)
(2)

(3)
No.

Name

(1)

Model

(2)

Model Selection(4)

Description
Displays current model code. By selecting (2) Model Selection, contents also changes.
*Direct editing cannot be done.
Determines model by selecting each selection. By selecting each selection, (1) Modelalso changes.

(3)

Next

For registering device in specified model, it moves to the confirmation dialog.

(4)

CancelIgnoring specified model, exit device registration wizard.

- 19 -

< 2. Name of device registration wizard (confirm) parts >Utilizing followings (1) Confirmation Contents, (2) Return, (3) Complete and (4) Cancel, it completes,
regresses or cancels device registration wizard.

(1)

(2)

No.

Name

(1)

Confirmation

(3)

(4)

Description
Registering model, device and series name are displayed.

Contents(2)

ReturnReturns to 2. Device registration wizard (setting of each device).

(3)

Complete

Registers the device as confirmed contents.

(4)

CancelIgnoring confirmed contents, exit device registration wizard.

- 20 -

5-2-2-3. Device Scan Dialog

It provides function of registration or setting from device registration window; model code scan ofconnecting device.

< 1. Name of device scan dialog parts >Utilizing followings (1) Communication Method, (2) Communication Condition Setting and (3) Scan, it
performs device scan.

(1)

(2)
(3)

No.

Name

(1)

Communication

Select communication methods to the connecting device from serial

Methodcommunication, USB communication or Ethernet communication.

Communication

Set specifics selected at (1) Communication Method Selection.

<2.>

Starts device scan under the condition setting set above.

<3.>

(2)

Description

Reference-

Condition
Setting(3)

Scan- 21 -

< 2. Name of device scan dialog; Serial communication condition setting parts >Utilizing followings (1) Port, (2) Device Address Range, (3) Retry Count and (4) Time Out, it sets underthese conditions.

(2)

(1)
(3)
No.

(4)
Name

Description

(1)

Port

Select arbitrary port from the list of port set on 6-2-1. Port Registration Window.

(2)

Device Address

Specify desirable device address range to scan in a range of 1 to 99.

Range
(3)

Retry Count

Specify communication retry count at scan in a range of 0 to 50.

(4)Time OutSpecify communication time out at scan in a range of 0 to 10000.

- 22 -

< 3. Name of device scan dialog; dialog after the scanning parts >Utilizing following functions (1) List of Scan Result, (2) Add Button, (3) Setting Button, (4) Web Connection
Button and (5) FTP Connection Button, it registers or sets device information.

(1)
(2)

(3)

(4)

(5)

No.

Name

Description(1)

List of Scan

After scan of connecting device, displays address which model code has been read andResultlist of device model code.
*Display is blank before the scan.

(2)

Add Button(3)

Setting Button

Adds device information to the list of device registration by utilizing device model code oftarget row.
Startups parameter setting dialog for the device found by device scan.
*Refer to 6-3-1. Parameter Setting for details.

(4)
(5)

Web Connection

By utilizing standard browser, displays Web setting dialog for target device.

Button*Operation is available only if the target device supports Web setting.

FTP Connection

By utilizing standard browser, displays FTP setting dialog for target device.

Button*Operation is available only if the target device supports FTP setting.

- 23 -

5-2-2-4. Device Setting Edit Dialog

It provides editing functions of device setting information.

< Name of device setting edit dialog parts >Utilizing following functions (1) Name, (2) Retry Count, (3) Time Out [ms], (4) Command Interval [ms],(5) Delay between Devices [ms], (6) Updata Interval [ms], (7) Serial Communication Setting, (8) OK and(9) Cancel, it edits or cancels device setting information.

(1)
(2)

(3)

(4)

(5)

(6)

(7)
(8)

No.

Name

(9)

Description

(1)

Name

Specify device name to display. Up to 30 characters are able to be entered.

(2)

Retry Count

Specify retry count at communication in a range of 0 to 50.

(3)Time Out [ms]Specify response time out [ms] at communication in a range of 0 to 10000.

(4)Command Interval [ms] Specify sending command interval [ms] at communication in a range of 0 to 10000.

(5)

Delay between

Specify delay between devices [ms] at acquisition in a range of 0 to 10000.

Devices [ms](6)

Update Interval [ms]Specify update interval [ms] at communication in a range of 0 to 3600.

(7)

Serial Communication

Specify device address for serial communication in a range of 1 to 99 and select andSetting

specify a port from the ports registered at 6-2-1.Port Registration Window.

OK

Retains setting information on the device setting edit dialog and reflects setting(8)

contents on device registration window.
(9)

CancelIgnores setting information on the device setting edit dialog and cancels reflecting
setting contents on device registration window.

- 24 -

5-2-3. Summary Display Window
It provides function which accumulates fragmental process information in the application and display as alist.

< Name of summary display window parts >Utilizing following functions (1) Export, (2) Summary Search, (3) Setting, (4) Delete, (5) Print and (6) List of
Summary Display, it displays summary information for browsing.

(1) (2) (3) (4) (5)

(6)

No.
(1)

Name
ExportDescription
Output currently displaying list of summary as a file.

Reference-*Select file format from CSV, Text or Excel.
(2)

Summary

Displays summary search dialog. Search specified summary information by

Searchentering search condition.

(3)

Setting

Performs various settings for summary display window.

(4)

DeleteDelete selected summary information from displayed summary.

5-2-3-1
5-2-3-2-*Deleting multiples is available.
(5)

Print

Output currently displayed list of summary as a print.

-(6)

List of Summary

Accumulates and displays fragmental process information in the-

Display

application.
Sort filter function is at header of each column. By utilizing filter window
shown below, specify desired event type to display.

Filter can be switched with orwithout by check box operation.

*With check mark: Displays/Without check mark: Hidden*Right click sub menu: Resets sort. Clicking the button resets sort status.

- 25 -

5-2-3-1. Summary Search Dialog

It provides search function by specifying multiple search conditions and search from list of summary.

< 1. Name of summary search dialog parts >Utilizing followings (1) Event Type Selection, (2) Start Point Date, (3) End Point Date, (4) Category,(5) Group Name, (6) Contents, (7) Comment, (8) Search All, (9) Search Previous and (10) Search Next, it
perform search process for list of summary.
*For specifying multiple condition except for event type, all are searched by AND condition.

(2)
(1)

(3)
(4)

(5)

(6)

(7)
(8)

(9)

No.

Name

Description(1)

Even TypeSpecify event type. All three types; information, alarm and abnormal are

Selection

specified as default value. If searching only for limited event information,(10)
Reference-

check only desired event type.
(2)
(3)
(4)

Start Point

Specify the start point date. By checking the checkbox enables the

Datefunction.

End Point

Specify the end point date. By checking the checkbox enables the

Datefunction.

Category

Specify summary event category from selection or input directly.

-

By checking the checkbox enables the function.
(5)

Group Name

Specify group name by input directly.

-

By checking the checkbox enables the function.
(6)

Contents

Specify contents by input directly.

-

By checking the checkbox enables the function.
(7)

Comment

Specify comments by input directly.

-

By checking the checkbox enables the function.
(8)

Search All

Searches all the information matches to specified condition from the list of<2.>summary and displays in list of result.
(9)
(10)

SearchSearch information matches to specified condition from the list ofPrevioussummary upward.

Search Next

Search information matches to specified condition from the list ofsummary downward.

- 26 --

< 2. Name of summary search dialog (list of search result display) parts >Utilizing followings (1) List of Summary Result Display, it perform search process for list of summary.

(1)

No.
(1)

Name

Description

List of

Displays all the search result matches to the search condition. Double clicking summary

Summary

information on the list of search result highlights target summary information on the

Result Display

summary display window.
*Before clicking Search All, list of search result is hidden.

- 27 -

5-2-3-2. Summary Setting Dialog

It provides setting functions of summary display window; list of summary display.

< Name of summary setting dialog parts >Utilizing followings (1) Buffer Sample Count, (2) Font, (3) Information Row Setting, (4) Alarm Row Setting,(5) Abnormal Row Setting, (6) OK and (7) Cancel, it perform setting process for list of summary display.

(1)

(2)
(3)

(4)

(5)

(6)

(7)

No.

Name

Description(1)

Buffer Sample Count

Specify summary information count which this application retains in a range of 50 to
5000.
Summary information in the range of specified count is kept although application exit
and the information can be browsed at next startup.

(2)

Font

Specify font information of summary display window.

(3)

Information Row

Specify summary font color and background color for information event.

Setting(4)

Alarm Row SettingSpecify summary font color and background color for alarm event.

(5)

Abnormal Row

Specify summary font color and background color for abnormal event.

Setting(6)

OK

Retains setting information on the summary setting dialog and reflects setting
contents on the summary display window.

(7)

CancelIgnores setting information on the summary setting dialog and cancels reflecting
setting contents on the summary display window.

- 28 -

5-2-4. Version Information Dialog
It provides function of displaying list of version information for this application and device DLL which thisapplication retains.

< Name of version information dialog parts >Utilizing followings (1) Basic Application Information, (2) DLL Information for Each Function, (3) RegisteredDevice DLL Information and (4) OK, it displays version information.

(1)

(2)

(3)

(4)

No.
(1)

Name
Basic Application

Description
Displays main EXE version information of this application.

Information(2)

DLL Information for

Displays DLL version information for each function used in this application.

Each Function(3)

Registered Device

Displays DLL version information which have connecting device information used in

DLL Information

this application.
*Information varies depending on the extension of registered DLL.

(4)

OK

Close the version information dialog.

- 29 -

5-3. Operation of Parameter Setting
It provides function of parameter setting for the device registered at 6-2-2.
Device Registration Window.

< Name of parameter setting at launcher >Utilizing following (1) Registered Device Setting, (2) Parameter File Setting (file start) and (3) Parameter
File Setting (model specified start) it sets each parameter of registered device.
*Refer to the section 3-1 List of function restriction of free of charge edition about restriction of free of charge
edition.

(1)

(2)

(3)

No.

Name

Description

Reference(1)

Registered

List of registered device is displayed by clicking on the dropdown button.

DeviceSelect desired registered device to set from the list of registered device.

5-3-1

Setting

Click on the desired device to set.

Displays parameter setting window.
*Contents of setting item differ depending on the specified device.
(2)

Parameter

File open dialog is displayed to search for parameter file etc.

File Setting

Display parameter setting window (off-line) by using specified parameter
file.

(file start)

5-3-2*Contents of setting item differ depending on the specified device.
(3)

Parameter

Device selection wizard starts to specify a device to set parameter file.

File Setting

Display parameter setting window (off-line) by using specified device
model specified on selection method.

(modelspecified*Contents of setting item differ depending on the specified device.

start)*Device selection wizard is similar to 5-2-2-2. Device Registration Wizard- 30 -5-3-2

5-3-1. Parameter Setting Window
It provides parameter setting function for registered device.
At startup of the window, it starts up reading setting function for the registered device. If reading is
succeeded, current setting value is reflected on the window.

< Name of parameter setting window parts >Utilizing followings (1) Registered Device Information, (2) New, (3) Open File (4) Save As, (5) Copy,(6) Cut, (7) Paste, (8) Write Settings, (9) Read Settings, (10) Print, (11) Print List of Setting Information(Device dedicated format), (12) CSV Output List of Setting Information (Device dedicated format), (13)List of Setting Value Category and (14) Setting Value Input, it sets parameter for registered device.

(1)(2) (3) (4) (5) (6) (7) (8) (9) (10) (11) (12)

(13)
(14)

No.

Name

Description

Reference(1)

RegisteredDisplays display name and model code of registered device.

-

Expand new setting file on the window.

-

DeviceInformation
(2)

New

*Setting contents on the window will be initialized, perform (4) Save As if
it’s necessary.
(3)Open File

Specifies device parameter setting file (*.pasconf ) and expand on the-window.
*Confirmation message asking whether or not to rewrite registereddevice model is displayed when setting file which have different model
code is read.
Select "Yes" to rewrite and "No" to leave as it is. (Setting file is read only
if "Yes" is selected.)(4)

Save As(5)

Copy

Specifies file name and save current setting contents as device-

parameter setting file (*.passconf ) format.
Copy selected cell information on the (12) Setting Value Input.

-*Shortcut is available with Ctrl + C key.
(6)

Cut

Cut selected cell information on the (12) Setting Value Input.
*Shortcut is available with Ctrl + X key.

- 31 -

-

(7)

Paste

Paste retained cell information by (5) Copy or (6) Cut after the selected

-

cell.
*If data type of the cell is different, it cannot be pasted.
*Shortcut is available with Ctrl + V key.
(8)

Write Settings

Write setting value checked on the (11) List of Setting Value Category to

5-3-1-1

the device.
(9)

Read Settings

Read setting value checked on the (11) List of Setting Value Category

5-3-1-1from the device.
(10)

Print

Print setting value checked on the (11) List of Setting Value Category.

(11)

Print List ofPrint list of setting information of target model device.

-

Setting
Information(Device
dedicated
format)(12)

CSV Output

Output list of setting information of target model device in CSV.

List of Setting
Information(Device
dedicated
format)(13)

List of Setting

Displays list of setting device category of target registered device as tree

Value

view.

Category

Each category has checkbox and with the check mark, it is included as

5-3-1-2

process target and without check mark, it is excluded from processtarget.
Selecting the target item expand setting value group on the (12) Setting
Value Input for each category.
(14)

Setting Value

Setting value group for item selected on the (11) List of Setting Value

Input

Category is expanded and displayed. Inputted or changed values in thispart are set as parameters by (8) Write Settings.
*Enable value to be set may vary depending on the model code or
relating peripheral setting value.

- 32 --

5-3-1-1. Flow of Parameter SettingThis section describes operation of communication with the device and reading/writing of the parametersettings from the parameter setting window startup as flow chart.

< Operation image of parameter setting window >At the startup of the window, it starts obtaining
setting value of connecting target device.
*If the Cancel is clicked, initial value is displayed.
For writing settings for the connecting

Reading isdevice, to prevent unexpected setting

succeededchange, make sure to read the setting of
the device first and write only changed
contents.

When expand read setting value,font color changes to blue → black.

Change input type K2 → R.
Changing the setting value changes
font color black → blue.

For writing operation.

- 33 -

Click Write Settings icon.

Click Yes.

Writing progress bar is displayed.

Writing process complete
message is displayed.
Click OK.

Set input type R for connecting device.
Writing succeeded setting value changes
font color blue → black.

- 34 -

5-3-1-2. List of Setting Value Category (DB600)

It provides function of displaying list of setting value category for communication target device andexpends setting parameter on the parameter setting window when specified category is selected.
Each category has checkbox so switch of including to process target is also available.

< Setting Value Category List – Each Parameter Name >Utilizing followings (1) Operation Information, (2) Execution Parameter [MODE0], (3) Constant value
Operation Parameter [MODE1], (4) Program Parameter [MODE2], (5) Group Parameter [MODE3], (6)Event Paramete r[MODE4], (7) Input Parameter [MODE5], (8) Control Parameter [MODE6], (9) Output 2
Parameter [MODE7], (10) Communication Parameter [MODEa], (11) Transmission Output Parameter[MODEb], (12) Remote SV Input Parameter [MODEc], (13) External Input Parameter [MODEd] and (14)System Parameter [MODEt], it sets for category.

(1)
(3)
(5)
(7)
(9)
(11)
(13)

(2)
(4)
(6)
(8)
(10)
(12)
(14)

No.

Name

Description(1)

Operation

Displays operation information category.

Information

When specified, expand setting parameter at the setting value input.

Execution

Displays execution parameter category.

Parameter

When specified, expand setting parameter at the setting value input.

(2)

Reference
This section (a)This section (b)[MODE0](3)

Constant Value Displays constant value operation parameter category.
Operation

This section (c)When specified, expand setting parameter at the setting value input.

Parameter[MODE1](4)

Program

Displays program parameter category.

Parameter

When specified, expand setting parameter at the setting value input.

This section (d)[MODE2](5)Group

Displays group parameter category.

Parameter

When specified, expand setting parameter at the setting value input.

This section (e)[MODE3](6)

Event

Displays event parameter category.

Parameter

When specified, expand setting parameter at the setting value inputThis section (f)[MODE4](7)(8)

Input Parameter Displays input parameter category.
[MODE5]

When specified, expand setting parameter at the setting value input.

Control

Displays control parameter category.

Parameter

When specified, expand setting parameter at the setting value input.

[MODE6]- 35 -

This section (g)This section (h)(9)Output 2

Displays output 2 parameter category.

Parameter

When specified, expand setting parameter at the setting value inputThis section (i)[MODE7](10)

Communication

Displays communication parameter category.

Parameter

When specified, expand setting parameter at the setting value inputThis section (j)[MODEa](11)

Transmission

Displays transmission parameter category.

OutputWhen specified, expand setting parameter at the setting value inputThis section (k)Parameter[MODEd](12)

Remote SV

Displays remote SV input parameter category.

This section (l)Input Parameter When specified, expand setting parameter at the setting value input[MODEc](13)

External Input

Displays external input parameter category.

Parameter

When specified, expand setting parameter at the setting value inputThis section (m)[MODEd](14)

System

Displays system category.

This section (n)Parameter

When specified, expand setting parameter at the setting value input.

[MODEt]- 36 -

(a) Operation Information

It provides function of editing operation information.

< Name of operation information each setting value >Utilizing following various setting values (1) Manual Output, (2) Manual Output [Second], (3) A/M Switch(4) Alarm Reset, (5) Auot Tuning, (6) FB Tuning, (7) Run/Ready Switch, (8) Constant Value Operation
Execution No., (9) Execution Pattern No. and (10) Program Drive, it edits operation information.

(1)

(2)

(3)

(4)

(5)

(6)

(7)

(8)

(9)
No.

(10)
Name

Description(1)

Manual Output

Specify manual output value in the rage of -5 to 105.
*Writing is available only if MANUAL (manual output operation) is specified at (3) A/M
Switch(2)Manual Output

Specify manual output value [second] in the range of -5 to 105.

[Second]*Writing is available only if MANUAL (manual output operation) is specified at (3) A/M
Switch*Writing is available only if output 2 specification is with the connecting device.

(3)

A/M Switch

Select and specify Auto/Manual switch from AUTO (auto output operation) or

MANUAL (manual output operation).
*It is available for both output 1 and output 2.
(4)

Alarm Reset

Select and specify alarm reset setting form NON or RESET (alarm reset).
*Immediately after writing of RESET (alarm reset), it is switched to NON.

(5)

Auto Tuning

Select and specify auto tuning form END (auto tuning end) or START (auto tuning start).
*If auto tuning of the connecting device is other than END (auto tuning end), writing of
START (auto tuning start) is not available.
* Write END (auto tuning end), if force quitting during auto tuning.
*If Program operation RESET is specified at 810) Program Drive, writing is not available(except for constant value operation).

(6)

FB Tuning

Select and specify FB tuning setting from END (FB tuning end) or START (FB tuning
start).
*Writing START (FB tuning) is available, only if program operation RESET is specified in(10) Program Drive.
*If auto tuning or FB tuning of connecting device is executing, writing START (autotuning start) is not available.
*Writing is available only if ON-OFF servo output type specification is with theconnecting device.

(7)

Run/ReadySelect and specify Run/Ready switch from RUN status or READY status.

Switch*If READY (constant value operation RUN/READY switch) is specified at the section (m)External Input Parameter- External Signal Input Assignment, writing is not available.

- 37 -

(8)Constant Value
Operation
execution No.

Select and specify constant value operation execution No. from execution No.1,
execution No.2, execution No.3, execution No.4, execution No.5, execution No.6,
execution No.7 or execution No.8.
*If SEL1 (Execution No. Selection 1_bit0), SEL2 (Execution No.Selection2_bit1), SEL4
(Execution No. Selection 3_bit2) or SEL8 (Execution No. Selection 8_bit3) is specified at
the section (m) External Input Parameter- External Signal Input Assignment, writing is
not available.

(9)

Execution PatternNo.

Select and Specify execution No. from Pattern 1, Pattern 2, Pattern 3 or Pattern 4.
*If specified value in the (10) Program Drive is other than program operation RESET,
writing is not available.
*If specified value in the section (d) Program Parameter – general program/constantvalue operation switch is constant value operation, writing is not available.

(10)

Program Drive

Select and specify program drive setting from RESET (program operation RESET), RUN(program operation RUN), STOP (program operation STOP) or ADV (program operation
ADVANCE).
*If specified value in the section (d) Program Parameter - general, program/constantvalue operation switch is constant value operation, writing is not available.
*If RESET (program operation RESET) of connecting device is executing, writing of STOP
(program operation STOP) and ADV (program operation ADVANCE) is not available.
*If END (program operation END) of connecting device is executing, writing of RUN (program
operation RUN), STOP (program operation STOP) and ADV (program operation ADVANCE) is
not available.

- 38 -

(b) Execution Parameter [MODE0]

It provides function of editing execution parameter setting.

< Name of execution parameter each setting value >Utilizing following various setting values (1) SV, (2) Step Time “hh” or “mm”, (3) Step Time “mm” or “ss”,(4) Event 1 (5) Event 2, (6) Event 3, (7) Event 4, (8) P, (9) I, (10) D, (11) Output Limiter Lower Limit, (12)Output Limiter Upper Limit and (13) Guaranty Soak, it edits execution parameter.

(1)

(2)

(3)

(4)

(5)

(6)

(7)

(8)

(9)

(10)

(11)

(12)

(13)
No.

Name

Description

(1)

SV

Specify SV value in the range of Input Parameter SV limiter upper/lower limit in thesection (g).
*Decimal point position interlocks with SV decimal point in the section (g) Input
Parameter.

(2)

Step Time “hh” orSpecify step time “hh” or “mm” in the range of 0 to 999.

“mm”

*Time unit “hh” or “mm” interlocks with time unit of the section (d) Program
Parameter – general.
*If specified value in the section (d) Program Parameter - status, program/constantvalue operation switch is constant value operation, writing is not available.

(3)

Step Time “mm” orSpecify step time “mm” or “ss” in the range of 0 to 59.

“ss”*Time unit “hh” or “mm” interlocks with time unit of the section (d) Program
Parameter – general.
*If specified value in the section (d) Program Parameter - status, program/constantvalue operation switch is constant value operation, writing is not available.

- 39 -

(4)Event 1

Specify event value 1 to 4 in the range of -19999 to 30000.

(5)Event 2

*Decimal point position is specified as the section (f) Event Parameter each event(6)

Event 3mode as follows.

(7)

Event 4【
PV.H(Absolute value higher limit alarm)/PV.L(Absolute value lower limit alarm)/
DV.H(Deviation higher limit alarm)/DV.L(Deviation lower limit alarm)/
ADV.H(Absolute value deviation higher limit alarm)/ADV.L(Absolute value deviation
lower limit alarm)/
SV.H(Setting value higher limit alarm)/SV.L(Setting value lower limit alarm)】
Interlocks with SV decimal point.
【MV.H(Output value higher limit alarm)/MV.L(Output value lower limit alarm)/
CT.H(Heater higher limit alarm)/CT.L(Heater lower limit alarm)】
Specifies 1 digit.
【TIME1(Timer 1 alarm)/TIME2(Timer 2 alarm)】
No decimal point.
*If the section (f) Event Parameter event mode is FAIL (FAIL alarm) or status event,writing is not available.

(8)

P

Specify PID ”P” in the range of 0 to 999.9.
*0=Two position control(9)ISpecify PID ”I” in the range of 0 to 9999.
*0=∞

(10)

D

Specify PID ”D” in the range of 0 to 9999.
*0=Calculation OFF(11)

Output Limiter Lower

Specify output limiter lower limit in the range of -5 to 100.

Limit

*If specified value is not in the proper relation with (12) Output Limiter Upper Limit,
writing is not available.

(12)

Output Limiter Upper
Limit

Specify output limiter upper limit in the range of 0 to 105.
*If specified value is not in the proper relation with (11) Output Limiter Lower Limit,
writing is not available.

(13)

Guaranty SoakSpecify guaranty soak value in the range of 0 to 30000.
*0=Function OFF*Decimal point position interlocks with SV decimal point in the section (g) Input
Parameter.
*If specified value in the section (d) Execution Parameter - general,
program/constant value operation switch is constant value operation, writing is notavailable.

- 40 -

(c) Constant Value Operation Parameter [MODE1]

It provides function of editing constant value operation parameter setting.

< Name of constant value operation parameter each setting value >Utilizing following various setting values (1) Constant Value Operation Execution No, (2) Auto Tuning, (3)A/M Switch, (4) Manual Output, (5) Manual Output [Second], (6) PV Start, (7) Program/Constant Value
Operation Switch, (8) Run/Ready Switch, (9) Alarm Output at Ready and (10) Alarm Reset, it edits
constant value operation parameter status.

(1)

(2)

(3)

(4)

(5)
No.

Name

(1)

SV Rate-of-Change
Fall

(2)

SV Rate-of-Change
Rise

(3)

SV Rate-of-Change

Description
Specify SV rate-of-change fall in the range of -1999.9 to 0.
*Decimal point position interlocks with the section (g) Input Parameter SV decimal
point.
Specify SV rate-of-change rise in the range of 0 to 2000.
*Decimal point position interlocks with the section (g) Input Parameter SV decimal
point.
Specify SV rate-of-change time unit from S (second), M (minute) and H (hour).

Time Unit(4)

PV Start

Select and specify PV start setting from SV start or PV start.

(5)

Alarm Output atSelect and specify alarm output at Ready setting from OFF (alarm event calculation

Readydisable) or ON (alarm event calculation enable).

- 41 -

(d) Program Parameter [MODE2]

It provides function of editing constant value operation parameter setting.

< Program parameter setting image >Graph is made by inputting setting value.

< Name of program parameter – general, each setting value >Utilizing following various setting values (1) Program/Constant Value Operation Switch, (2) Time Unit, (3)Pattern Repeat, (4) Operation Screen Time Display, (5) Pattern Clear, (6) Pattern Copy Source, and (7)Pattern Copy Destination, it edits program parameter - general.

(1)

(2)

(3)

(4)

(5)

(6)

(7)
No.
(1)

(2)

Name

Description

Program /ConstantSelect and specify program/constant value operation switch form program

Value Operation Switch

operation or constant value operation.
*Writing is available only if program specification is with the connecting device.

Time Unit

Select and specify time unit setting from H:M (hour: minute) or M:S (minute:second).
*If specified value in the (a) Operation Information, Program Drive is other thanprogram operation RESET, writing is not available.

(3)

Pattern Repeat

Specify pattern repeat setting in the range of 0 to 9999.

- 42 -

(4)

(5)

(6)

Operation Screen Time

Select and specify operation screen time display setting from Step elapsed time or

Display

Step remaining time.
*Writing is available only if program function specification is with the connecting
device.
Select and specify pattern clear setting from Pattern 1, Pattern 2, Pattern 3, Pattern
4 or All patterns.
*If specified value in the (a) Operation Information, Program Drive is other thanPattern Clear

Pattern Copy Source

program operation RESET, writing is not available.
*If writing for this parameter, make sure not to write pattern 1 to 4.
Select and Specify pattern copy source from Pattern 1, Pattern 2, Pattern 3 orPattern 4.
*If copy source pattern is not set (number of steps is 0), writing is not available.
*If copy destination is not cleared (number of steps is other than 0), writing is notavailable*If the pattern of copy source and pattern of copy destination is the same, writing is
not available.
*Clear the copy destination pattern by (5) Pattern Clear then execute pattern copy.

(7)

Pattern Copy
Destination*If writing for this parameter, make sure not to write pattern 1 to 4.
*If specified value in the (a) Operation information, Program Drive is other thanprogram operation RESET, writing is not available.
Select and specify pattern clear setting from Pattern 1, Pattern 2, Pattern 3, Pattern
4 or All patterns.
*If copy source pattern is not set (number of steps is 0), writing is not available.
*If copy destination is not cleared (number of steps is other than 0), writing is notavailable*If the pattern of copy source and pattern of copy destination is the same, writing is
not available.
*Clear the copy destination pattern by (5) Pattern Clear then execute pattern copy.
*If writing for this parameter, make sure not to write pattern 1 to 4.
*If specified value in the (a) Operation information, Program Drive is other thanprogram operation RESET, writing is not available.

- 43 -

< Name of program parameter, pattern 1 to 4 setting – general each setting value>
Utilizing following various setting values (1) Number of Steps, (2) PV Start, (3) Start SV, (4) Output 1 at
END, (5) Output 2 at END and (6) Linked to Pattern, it edits program parameter setting - general.

(1)

(2)

(3)

(4)

(5)

(6)

No.

Name

(1)

Number of Steps

Description
Specify number of steps in the range of 0 to 12.
*Specifying number of steps 1 or more enables step setting.

(2)

PV Start

Select and specify PV start setting from SV start or PV start.

(3)

Start SV

Specify start SV setting value in the range of the section (g) Input Parameter SV limit.
*Decimal point position interlocks with SV decimal point in the section (g) Input
Parameter.

(4)

Output 1 at End

Specify output 1 at End setting to CONTROL or in the range of -5 to 105.

(5)

Output 2 at End

Specify output 2 at End setting to CONTROL or in the range of -5 to 105.

(6)

Linked to Pattern*Writing is available only if output 2 specification is with the connecting device.
Select and specify link to pattern setting from No pattern link, Link to pattern 1, Link topattern 2, Link to pattern 3 or Link to pattern 4.

< Name of program parameter, pattern 1 to 4 setting – step setting each setting value >Utilizing following various setting values (1) SV., (2) Time “hour” or “minute”, (3) Time “minute” or “second”
and (4) Execution No. of 8 Types, it edits program parameter setting - step setting of each No.

(1)

(2)

(3)

(4)

No.

Name

Description

(1)

SV

Specify SV setting value of target No. in the range of 5-3-1-2-1. Input parameter SV
limit.

(2)

Time “hour” orSpecify time “hour” or “minute” of target No.in the range of 0 to 999.

“minute”(3)

Time “minute” orSpecify time “minute” or “second” of target No.in the range of 0 to 59.

“second”(4)

Execution No. of 8

Specify execution No. of 8 types in the range of 0 to 8.

Types*If 0 is specified, it becomes continuous No. of previous step.

- 44 -

(e) Group Parameter [MODE3]

It provides function of editing group parameter setting.

< Name of group parameter each setting value >Utilizing following various setting values (1) SV Value, (2) Event 1, (3) Event 2, (4) Event 3, (5) Event 4,(6) P, (7) I, (8) D, (9) Output Limiter Lower Limit (10) Output Limiter Upper Limit (11) Guaranty Soak, itedits 8 types of parameter.

(1)
(3)
(5)
(7)
(9)

(2)
(4)
(6)
(8)

(10)
(11)
No.

Name

Description(1)

SV Value

Specify SV value in the range of (g) Input Parameter SV limiter upper/lower limit.
*Decimal point position interlocks with SV decimal point in the section (g) Input
Parameter.

(2)

Event 1

Specify event value 1 to 4 in the range of -19999 to 30000.

(3)

Event 2

*Decimal point position is specified as the section (f) Event Parameter each event mode

(4)

Event 3

as follows.

(5)

Event 4【
PV.H(Absolute value higher limit alarm)/PV.L(Absolute value lower limit alarm)/
DV.H(Deviation higher limit alarm)/DV.L(Deviation lower limit alarm)/
ADV.H(Absolute value deviation higher limit alarm)/ADV.L(Absolute value deviation
lower limit alarm)/
SV.H(Setting value higher limit alarm)/SV.L(Setting value lower limit alarm)】
Interlocks with SV decimal point.
【MV.H(Output value higher limit alarm)/MV.L(Output value lower limit alarm)/
CT.H(Heater higher limit alarm)/CT.L(Heater lower limit alarm)】
Specifies 1 digit.
【TIME1(Timer 1 alarm)/TIME2(Timer 2 alarm)】
No decimal point.
*If the section (f) Event Parameter event mode is FAIL (FAIL alarm) or status event,writing is not available.

(6)

P

Specify PID ”P” in the range of 0 to 999.9.
*0=Two position control(7)

ISpecify PID ”I” in the range of 0 to 9999.
*0=∞

(8)

D

Specify PID ”D” in the range of 0 to 9999.
*0=Calculation OFF- 45 -

(9)Output Limiter

Specify output limiter lower limit in the range of -5 to 100.

Lower Limit*If specified value is not in the proper relation with (10) Output Limiter Upper Limit,
writing is not available.

(10)Output Limiter

Specify output limiter upper limit in the range of 0 to 105.

Upper Limit,*If specified value is not in the proper relation with (9) Output Limiter Lower Limit, writing
is not available.

(11)

Guaranty SoakSpecify guaranty soak value in the range of 0 to 30000.
*0=Function OFF*Decimal point position interlocks with SV decimal point in the section (g) Input
Parameter.
*If specified value in the section (d) Program Parameter - general, program/constantvalue operation switch is constant value operation, writing is not available.

- 46 -

(f) Event Parameter [MODE4]

It provides function of editing event parameter.

< Name of event parameter each setting value >Utilizing following various setting values (1) Event No., (2) Expansion Function, (3) Event Mode, (4) Event
Output Phase and (5) Event Deadband, it edits event parameter.

(1)

(2)

No.

Name

(3)

(4)

(5)

Description(1)

Event No.

Displays EV.1 to EV.9 setting header. Changing is not available.

(2)

Expansion Function

Select and specify event expansion function from none, with standby, with keep orwith keep+standby.
*Setting is only available for EV.1 to EV.4.

(3)

Event Mode

Select and specify fromNON (No event),
PV.H (Absolute value upper limit alarm), PV.L (Absolute value lower limit alarm),
DV.H (Deviation upper limit alarm), DV.L (Deviation lower limit alarm),
ADV.H (Absolute value deviation upper limit alarm),
ADV.L (Absolute value deviation lower limit alarm),
SV.H (Setting value upper limit alarm), SV.L (Setting value lower limit alarm),
MV.H (Output value upper limit alarm), MV.L (Output value lower limit alarm),
CT.H (Heater upper limit alarm), CT.L (Heater lower limit alarm),
TIME1 (Timer 1 alarm), TIME2 (Timer 2 alarm),
FAIL (FAIL alarm), C.RUN (Constant value operation RUN),
PRST.M (Preset manual status), REM (Remote input status),
SV.UP (SV rising status), SV.DW (SV falling status),
STEP (Step switch of program operation),
STEP1 (Step 1 of Program operation),
STEP2 (Step 2 of Program operation),
STEP3 (Step 3 of Program operation),
STEP4 (Step 4 of Program operation),
STEP5 (Step 5 of Program operation),
STEP6 (Step 6 of Program operation),
STEP7 (Step 7 of Program operation),
STEP8 (Step 8 of Program operation),
STEP9 (Step 9 of Program operation),
STEP10 (Step 10 of Program operation),
STEP11 (Step 11 of Program operation),
STEP12 (Step 12 of Program operation),
P.RUN (Program operation RUN),
STOP (Program operation STOP),ADV (Program operation ADVANCE),RESET (Program operation RESET),END (Program operation END) or
CONST (SV keep status of program operation).
*For EV.5 to EV.9selection other than alarm even are able to be specified.

- 47 -

*For CT.H (Heater upper limit alarm) and CT.L (Heater lower limit alarm), writing isavailable only if CT specification is with the connecting device.
*For TIME1 (Timer 1 alarm) and TIME2 (Timer 2 alarm), writing is available only if DI
specification is with the connecting device.
For
STEP (Step switch of program operation),
STEP1 (Step 1 of Program operation),
STEP2 (Step 2 of Program operation),
STEP3 (Step 3 of Program operation),
STEP4 (Step 4 of Program operation),
STEP5 (Step 5 of Program operation),
STEP6 (Step 6 of Program operation),
STEP7 (Step 7 of Program operation),
STEP8 (Step 8 of Program operation),
STEP9 (Step 9 of Program operation),
STEP10 (Step 10 of Program operation),
STEP11 (Step 11 of Program operation),
STEP12 (Step 12 of Program operation),
P.RUN (Program operation RUN),
STOP (Program operation STOP),ADV (Program operation ADVANCE),RESET (Program operation RESET),END (Program operation END) or
CONST (SV keep status of program operation), writing is available only if program
function is with the connecting device.
(4)

Event Output Phase

Select and specify event output phase from output of normal phase or output ofreverse phase.
* For EV.1 to EV.2, if connecting device is with no event output specification, writing
is not available (output of normal phase is fixed).
*For EV.3 to EV.4, if connecting device is with no transmission signal output,
4-20mA, 0-1V, 0-10V specification, writing is not available (output of normal phase isfixed).
*For EV.5 to EV.9, if connecting device is with communication interface RS422A+Event output 5 points (open corrector output), Event output 5 points (opencorrector output) specifications, writing is not available (output of normal phase isfixed).

(5)

Event Deadband

Specify event deadband in the range of 0 to 200 (0 to 20000)*Decimal point position is specified as (3) Event Mode as follows.
【
PV.H (Absolute value upper limit alarm), PV.L (Absolute value lower limit alarm),
DV.H (Deviation upper limit alarm), DV.L (Deviation lower limit alarm),
ADV.H (Absolute value deviation upper limit alarm),
ADV.L (Absolute value deviation lower limit alarm),
SV.H (Setting value upper limit alarm), SV.L (Setting value lower limit alarm)】
SV decimal point (maximum 3 digits)+1.
【
MV.H (Output value upper limit alarm), MV.L (Output value lower limit alarm),
CT.H (Heater upper limit alarm), CT.L (Heater lower limit alarm)】
Specify 1 digit.
*If event parameter (3) Event Mode is TIME1 (Timer 1 alarm), TIME2 (Timer 2
alarm), FAIL (FAIL alarm) or Status Event, writing is not available.

- 48 -

(g) Input Parameter [MODE5]

It provides function of editing input parameter.

< Name of input parameter each setting value >Utilizing following various setting values (1) Input Type, (2) Measuring Range Lower Limit, (3) Measuring
Range Upper Limit (4) SV Decimal Point (Linear Scale Decimal Point), (5) Linear Scale Zero, (6) Linear
Scale Span, (7) Sensor Correction, (8) Digital Filter, (9) SV Limiter Lower Limit, (10) SV Limiter Upper Limitand (11) SV Decimal Point for Display, it edits input parameter.

(1)

(2)

(3)

(4)

(5)

(6)

(7)

(8)

(9)

No.

(10)
Name

Description(1)

Input Type

Select and specify device input type from B, R, S, N, K1, K2, E, J, T, U, L, W-WRe526, W-WRe26, Platinel II, PtRh40-20, Au-Pt, 20mV, 100mV, 5V, 10V,Pt100Ω1,
Pt100Ω2, JPt100Ω1, JPt100Ω2 or Pt50Ω.
*Writing input type to the connecting device interlocks with the other parameters andupdates them; therefor it is recommended to write independently if changing the
input type.

(2)

Measuring Range

Specify measuring range lower limit in the measuring range of (1) Input Type.

Lower Limit*If specified value is not in the proper relation with (3) Measuring Range Upper Limit,
writing is not available.

(3)

Measuring Range

Specify measuring range Upper limit in the measuring range of (1) Input Type.

Upper Limit*If specified value is not in the proper relation with (2) Measuring Range Lower Limit,
writing is not available.

(4)

(5)

SV Decimal Point

Specify SV decimal point and linear scale decimal point.

*Linear Scale*If (1) Input Type is belonging to thermocouple or resistance thermometer, this setting

Decimal Point

value is fixed value and specifying arbitrarily is no available.
*Writing SV decimal point to the connecting device interlocks with the otherparameter range and updates them; therefor it is recommended to write
independently if changing the SV decimal point.

Linear Scale Zero

Specify linear scale zero position.
*Decimal point position is interlocking with (4) SV decimal point.
*If (1) Input Type is other than ±20mV, ±100mV, ±5V or ±10V, writing is not available.

(6)

Linear Scale Span

Specify linear scale span.
*Decimal point position is interlocking with (4) SV decimal point.
*If (1) Input Type is other than ±20mV, ±100mV, ±5V or ±10V, writing is not available.

(7)

Sensor Correction

Specify sensor correction in the range of -19999 to 20000.
*Decimal point position is PV decimal point (fixed value of each (1) Input Type)+1(8)Digital Filter

Specify digital filter in the range of 0 to 99.9.
*0=OFF- 49 -

(9)SV Limiter Lower

Specify SV limiter lower limit.

Limit*Decimal point position interlocks with (4) SV Decimal Point.
*If specified value is not in the proper relation with (10) SV Limiter Upper Limit,
writing is not available.

(10)

SV Limiter Upper

Specify SV limiter Upper limit.

Limit*Decimal point position interlocks with (4) SV Decimal Point.
*If specified value is not in the proper relation with (9) SV Limiter Lower Limit, writing
is not available.

(11)

SV Decimal Point forSpecify PV and/or SV decimal point position for operation window from the range of

Display

0 to 4.

- 50 -

(h) Control Parameter [MODE6]

It provides function of editing control parameter.

< Name of control parameter each setting value >Utilizing following various setting values (1) Pulse Cycle., (2) FB Zero, (3) FB Span, (4) FB Deadband,(5)Event Deadband/Output Deadband, (6) Control Operation Direct/Reverse, (7) Output Preset, (8) Output
Variation Limiter Fall, (9) Output Variation Limiter Rise, (10) PV Abnormal Output, (11) Preset Manual, (12)Operation at power On, (13) FF Control Switch and (14)Control Algorithm, it edits control parameter.

(1)

(2)

(3)

(4)

(5)

(6)

(7)

(8)

(9)

(10)

(11)

(12)

(13)

(14)
No.

Name

(1)

Pulse Cycle

Description
Specify pulse cycle in the range of 1 to 180.
*Writing is available only if output 1 of the connecting device is ONE-OFF pulse/SSR
drive pulse output type.

(2)

FB ZeroSpecify FB zero position in the range of -5 to 104.9.
*If specified value is not in the proper relation with (3) FB Span, writing is not available.
*Writing is available only if ON-OFF servo output type specification is with theconnecting device.

(3)

FB SpanSpecify FB span in the range of -4.9 to 105.
*If specified value is not in the proper relation with (2) FB Zero, writing is not available.
*Writing is available only if ON-OFF servo output type specification is with theconnecting device.

(4)

FB Deadband

Specify FB deadband in the range of1 to 20.
*Writing is available only if ON-OFF servo output type specification is with theconnecting device.

(5)

PID

Specify PID deadband/output deadband in the range of 0.0 to 9.9.

Deadband/Output
Deadband(6)

Control Operation

Select and specify control operation form DIRECT(cooling control)/REVERSE(heating

Direct/Reverse

control).

(7)

Output Preset

Specify output preset value in the range of -100 to 100.

(8)Output Variation

Specify output variation limiter fall in the range of -100 to -0.1.

Limiter Fall

- 51 -

(9)Output Variation

Specify output variation limiter rise in the range of 0.1 to 100.

Limiter Rise
(10)

PV Abnormal

Specify PV abnormal output value in the range of -5 to 105.

Output(11)

Preset Manual

Specify preset manual value in the range of -5 to 105.

(12)

Operation at Power

Select and specify operation at power ON from Action just before the previous power
OFF status, Constant value operation READY status or Program operation RESET
status.
Select and specify FF control switch from OFF or ON.
*Writing is available only if Z control specification is with the connecting device.
Select and specify control algorithm setting from position type PID or velocity type PID.
*Writing is available only if PID control specification is with the connecting device.

ON
(13)

FF Control Switch(14)

Control Algorithm- 52 -

(i) Output 2 Parameter [MODE7]

It provides function of editing output 2 parameter.

< Name of communication/transmission parameter each setting value >Utilizing following various setting values (1) Control Method, (2) P, (3) I, (4) D, (5) Output Gap, (6) PID
Deadband/Output Deadband, (7) Control Operation Direct/Reverse, (8) Output Preset, (9) Split Direct,(10) Split Reverse, (11) Pulse Cycle, (12) Output Limiter Lower Limit, (13) Output Limiter Upper Limit,(14) Output Variation Limiter Fall, (15) Output Variation Limiter Rise and (16) Preset Manual, it edits output
2 parameter.

(1)

(2)

(3)

(4)

(5)

(6)

(7)

(8)

(9)

(10)

(11)

(12)

(13)

(14)

(15)

(16)

No.

Name

(1)

Control Method

Description
Select and specify control method from PID control or split control.
*Writing is available only if output 2 specification is with the connecting device.

(2)

P

Specify PID ”P” in the range of 0 to 999.9.
*0=Two position control*Writing is available only if output 2 specification is with the connecting device.

(3)

I

Specify PID ”I” in the range of 0 to 9999.
*0=∞*Writing is available only if output 2 specification is with the connecting device.

(4)

D

Specify PID ”D” in the range of 0 to 9999.
*0=Calculation OFF*Writing is available only if output 2 specification is with the connecting device.

(5)

Output Gap

Specify output gap in the range of -100.0 to 100.0.
*Writing is available only if output 2 specification is with the connecting device.

(6)

PID

Specify PID deadband/output deadband in the range of 0.0 to 9.9.

Deadband/Output*Writing is available only if output 2 specification is with the connecting device.

Deadband
Direct/Reverse

Select and specify control operation form DIRECT (cooling control)/REVERSE (heating
control).
*Writing is available only if output 2 specification is with the connecting device.

(8)

Output Preset

Specify output preset value in the range of -100 to 100.

(9)

Split Direct(7)

Control Operation*Writing is available only if output 2 specification is with the connecting device.
Specify split direct in the range of 0 to 60.
*Writing is available only if output 2 specification is with the connecting device.

- 53 -

(10)Split Reverse(11)

Pulse Cycle

Specify split reverse in the range of 40 to 100.
*Writing is available only if output 2 specification is with the connecting device.
Specify pulse cycle in the range of 1 to 180.
*Writing is available only if output 2 of the connecting device is ONE-OFF pulse/SSR drive
pulse output type.

(12)

Output Limiter

Specify the output limiter lower limit in the range of -5 to 100.

Lower Limit*Writing is available only if output 2 specification is with the connecting device.
*If specified value is not in the proper relation with (13) Output Limiter Upper Limit, writing
is not available.

(13)

Output Limiter

Specify output limiter upper limit in the range of 0 to 105.

Upper Limit*Writing is available only if output 2 specification is with the connecting device.
*If specified value is not in the proper relation with (12) Output Limiter Lower Limit, writing
is not available.

(14)
(15)
(16)

Output Variation

Specify output variation limiter fall in the range of -100 to -0.1.

Limiter Fall

*Writing is available only if output 2 specification is with the connecting device.

Output Variation

Specify output variation limiter rise in the range of 0.1 to 100.

Limiter Rise

*Writing is available only if output 2 specification is with the connecting device.

Preset Manual

Specify preset manual value in the range of -5 to 105.
*Writing is available only if output 2 specification is with the connecting device.

- 54 -

(j) Communication Parameter [MODEa]

It provides function of editing communication parameter.

< Name of 8 types of parameter each setting value >Utilizing following setting value (1) Digital Transmission Type, (2) [COM1] Communication protocol, (3)[COM1] Communication Function, (4) [COM1] Instrument No., (5) [COM1] Communication Transmission
Rate, (6) [COM1] Communication Character, (7) [COM1] Digital Transmission Type, (8) [COM1] Digital
Transmission cycle, and (9) [COM1] Communication Display, it edits transmission output parameter.
*(2) to (9) parameters are reading only.

(1)

(2)

(3)

(4)

(5)

(6)

(7)

(8)

(9)
No.
(1)

Name

Description

Digital

Select and specify digital transmission type setting form SV (setting value), PV (measured

Transmission

value), MV (output value), MV2 (output value 2) or MFB (analog FB transmission).

Type*For MV2 (output value 2), writing is available only if output 2specification is with theconnection device.
*For MFB (analog FB transmission), writing is available only if ON-OFF servo output typespecification is with the connecting device.
*Writing is available only if communication interface specification is with the connecting
device.

(2)

(3)

(4)
(5)

[COM1]

Select and specify [COM1] communication protocol setting from MODBUS RTU mode,

Communication

MODBUS ASCII mode, or Private mode.

Protocol*This parameter is reading only. It reads and displays setting above from connection device.

[COM1]

Select and specify [COM1] communication function setting from host communication, digital

Communicationtransmission, or digital remote.

Function*This parameter is reading only. It reads and displays setting above from connection device.

[COM1]

Specify [COM1] instrument No. in the range of 1 to 99.

Instrument No.

*This parameter is reading only. It reads and displays setting above from connection device.

[COM1]

Select and specify [COM1] communication transmission rate setting from 2400bps,

Communication

4800bps, 9600bps, 19200bps, or 38400bps.

Transmission*This parameter is reading only. It reads and displays setting above from connection device.

Rate
(6)

[COM1]

Select and specify [COM1] communication character setting from 7-bit/ even/ stop 1, 7-bit/

Communicationeven/ stop 2, 7-bit/ odd/ stop 1, 7-bit/ odd/ stop 2, 8-bit/ none/ stop 1, 8-bit/ none/ stop 2, 8-

Character

bit/even/ stop 1, 8-bit/ even/ stop 2, 8-bit/odd/stop 1, or 8-bit/ odd/ stop 2.
*This parameter is reading only. It reads and displays setting above from connection device.

(7)[COM1]

Select and specify [COM1] digital transmission type setting from SV (setting value), PV

Digital(measured value), MV (output value), MV2 (output value 2), or MFB (analog FBTransmission

transmission).

Type*This parameter is reading only. It reads and displays setting above from connection device.

- 55 -

(8)[COM1]

Select and specify [COM1] digital transmission setting from 100msec interval, 200msec

Digitalinterval, or 1000msec interval.

Transmission*This parameter is reading only. It reads and displays setting above from connection device.

cycle
(9)

[COM1]

Select and specify [COM1] digital communication display setting from Hide, or Display.

Communication*This parameter is reading only. It reads and displays setting above from connection device.

Display- 56 -

(k) Transmission Output Parameter [MODEb]

It provides function of editing transmission output parameter.

< Name of feedback each setting value >Utilizing following various setting values (1) Transmission Type, (2) Transmission Scale Zero and (4)Transmission Scale Span, it edits communication and transmission output parameter.

(1)

(2)

(3)
No.

Name

Description(1)

Transmission

Select and specify transmission type setting form SV (setting value), PV (measured

Typevalue), MV (output value) or MV2 (output value 2).
*For MV2 (output value 2), writing is available only if output 2specification is with theconnection device.
*For MFB (analog FB transmission), writing is available only if ON-OFF servo output
type specification is with the connecting device.
*Writing is available only if transmission signal output specification is with the connecting
device.

(2)

Transmission

Specify transmission scale zero position in the range of -19999 to 30000.

Scale Zero

*Writing is available only if transmission signal output specification is with the connecting
device.
*If (1) Transmission Type is SV (setting value) or PV (measured value), decimal point
position is SV decimal point and MV (output value), MV2 (output value 2) or MFB(analog FB transmission), decimal point position is decimal point 1 digit.

(3)

Transmission

Specify transmission scale span in the range of -19999 to 30000.

Scale Span

*Writing is available only if transmission signal output specification is with the connecting
device.
*If (1) Transmission Type is SV (setting value) or PV (measured value), decimal point
position is SV decimal point and MV (output value) or MV2 (output value 2), decimal
point position is decimal point 1 digit.

- 57 -

(l) Remote SV Input Parameter [MODEc]

It provides function of editing remote SV input parameter.

< Names of program parameter – general each setting value >Utilizing following various setting values (1) Remote/Local, (2) Remote Shift, (3) Remote SV Filter, (4)Tracking Switch to Local, (5) Remote SV Scale Zero, (6) Remote SV Scale Span, and (7) Cascade Input γ(ratio) and (8) Cascade Input β (bias), it edits remote SV input parameter.

(1)

(2)

(3)

(4)

(5)

(6)

(7)

(8)

No.

Name

Description
Select and specify remote/local setting from LOCAL (local SV) or REMORT (remote SV).
*Writing is available only if remote signal input or communication interface specification
is with the connecting device.
Specify remote SV shift value in the range of -19999 to 20000.
*Decimal point is the section (g) Input Parameter SV decimal point (maximum 3
digits)+1.
*Writing is available only if remote signal input or communication interface specification
is with the connecting device.
Specify remote SV filter value in the range of 0 to 99.9.
*Writing is available only if remote signal input or communication interface specification
is with the connecting device.

(1)

Remote/Local(2)

Remote SV Shift(3)

Remote SV Filter(4)

Tracking at

Select and specify tracking setting value at local switching form tracking function disable

Switching to Localor tracking function enable.
*Writing is available only if remote signal input or communication interface specification
is with the connecting device.

(5)

Remote SV Scale
Zero

(6)

Remote SV Scale
Span

(7)

Cascade Input
γ (ratio)(8)

Cascade Input
β (bias)Specify remote SV scale zero position in the range of -19999 to 30000.
*Decimal point position interlocks with the section (g) Input Parameter SV decimal point.
*Writing is available only if remote signal input specification is with the connecting
device.
Specify remote SV scale span in the range of -19999 to 30000.
*Decimal point position interlocks with the section (g) Input Parameter SV decimal point.
*Writing is available only if remote signal input specification is with the connecting
device.
Specify cascade input value γ in the range of 0.00 to 1.00.
*Writing is available only if remote signal input specification is with the connecting
device.
Specify cascade input value β in the range of -99.9 to 100.0.
*Writing is available only if remote signal input specification is with the connecting
device.

- 58 -

(m) External Input Parameter [MODEd]

It provides function of editing external input parameter.

< Name of system - external signal input each setting value >Utilizing following various setting values (1) External Signal Input Assignment No.1 to 7, it edit system.

(1)

No.
(1)

Name

Description

External Signal

Select and specify external signal input assignment No.1 to 7 from

Input Assignment

NON (No assignment),

No.1 to 7

READY (Constant value operation RUN/READY switch),
MANUA (AUTO/Manual output switch),
PRAT.M (Preset manual),
TIMER1 (Timer 1),
TIMER2 (Timer 2),
AL.RST (Alarm event reset),
SEL1 (Execution No. Selection 1_bit0),
SEL2 (Execution No. Selection 2_bit1),
SEL4 (Execution No. Selection 4_bit2),
SEL8 (Execution No. Selection 8_bit3),
PROGR (Program/Constant value operation switch),
P.RUN (Program operation RUN/STOP),
P.ADV (Program operation ADVANCE),
P.REST (Program operation RESET),PTRN1 (Program pattern 1_bit0),
PTRN2 (Program pattern 2_bit1) or
PTRN4 (Program pattern 4_bit2)*Writing is available only if external signal input specification is with the connecting
device.
*Writing of
PROGR (Program/Constant value operation switch),
P.RUN (Program operation RUN/STOP),
P.ADV (Program operation ADVANCE), P.REST (Program operation RESET),PTRN1 (Program pattern 1_bit0), PTRN2 (Program pattern 2_bit1) and
PTRN4 (Program pattern 4_bit2) is available, only if program function specification iswith the connecting device.

- 59 -

(n) System Parameter [MODEt]

It provides function of editing system parameter.

< Name of system - external signal input each setting value >Utilizing following various setting values (1) Key Lock, (2) Operation Initial Screen, (3) Operation Screen
SV Display, (4) Parameter Storage, (5) Port switch, (6) Multi-output switch, (7) Calculation RESET at
READY, (8) ARW lower limit, and (9) ARW upper limit, it edits system parameter.

(1)

(2)

(3)

(4)

(5)

(6)

(7)

(8)

(9)
No.

Name

Description(1)

Key LockSelect and specify key lock setting form without key lock, key lock of operation
parameter or key lock of all parameter.

(2)

Operation Initial

Select and specify operation initial screen setting from PV/SV display, PV/Step time

Screendisplay, PV/Output 1 display or PV/Output 2 display.
*Writing of PV/Output 2 display is available only if output 2 specification is with theconnecting device.

(3)
(4)

Operation Screen SV

Select and specify operation screen SV display setting from SV display, Blank

Display

display or Multi display.

Parameter Storage

Select and specify parameter storage setting form ON (With Setting parameterstorage) or OFF (Without Setting parameter storage).
*If specified value in the (d) Program Parameter – general, program /constant value
operation switch is program operation and (a) Operation Information, program drive,
program operation is specified, writing is not available.

(5)

Port switch

Select and specify port switch setting from engineering port or IR port.
*This parameter is reading only. It reads and displays setting above from connection
device.

(6)

Multi-output switch

Select and specify multi-output switch setting from normal output, ON-OFF pulse
output, analog output, or SSR drive pulse output.
*Writing is available if connection device is combination of following models.
・Adjustment output 1…SSR drive pulse output type・Adjustment output 2…Current output type or Voltage output type・Event output 2 points…With EV2(7)

Calculation RESET at

Select and specify calculation RESET at Ready setting from continues PID

READY

calculation while Ready status, or RESET PID calculation while Ready.

(8)

ARW lower limit

Specify ARW lower limit in the range of -100.0 to 0.0.

(9)

ARW upper limit

Specify ARW upper limit in the range of 0.0 to 100.0.

- 60 -

5-3-2. Parameter Setting Window (Off-Line)It provides parameter setting function of editing device parameter file (dose not provide communication).
There are startups by reading device parameter file and from device model selection wizard.
Refer to 5-3-1-2. List of Setting Value Category (DB600) for how to edit.
< Image of parameter setting window (off-line) >- 61 -

5-3-2-1. Setting Flow of Device Parameter

In flow chart, it describes the operation (startup from parameter setting window (off-line)) of editing andwriting set values that are either read from a device or set values of device specified value that thisapplication has and then writing to a device that has same model as the edited values of the device.

< Operation image of parameter setting window (off-line) >(1) A case of editing set values from starting up parameter setting window and opening device parameter file

Click parameter setting window (file startup).

File open dialog is displayed.
Select desired device parameter file to edit.

Parameter setting window (off-line) of device model that is included in a target
file is displayed.

- 62 -

Select and change parameter and rewrite and save edited contents.

Device parameter file is saved.

- 63 -

(2) A case of editing set values of parameter setting window from device selection wizard

Click parameter setting window (startup device selection wizard).

Device selection wizard is displayed.
Select device model code to edit its setting information.

Click [Next].

Click [Finish].

- 64 -

Parameter setting window (off-line) matches for selected device model isdisplayed.

Select and change parameter and save edited contents as a device parameter file.

Device parameter file is saved.

- 65 -

(3) A case of writing saved device parameter file to a read device

Select parameter setting window of a desired device
from registered information to reflect contents ofdevice parameter file.

Importing set value from connecting target device starts
at the window startup.
*Initial value is displayed if Cancel is clicked.
Importing complete.

Click [File open] button.

- 66 -

File open dialog is displayed.
Select the device parameter file that is just saved
before.

Present set contents aredeleted and message
addressing rewrite isdisplayed.
Click [ Yes ].

If format of registered device and selected device
parameter file are different, message addressing
registered device format change is displayed.
Click [ Yes ].
*It is not displayed if device format matches.
Click [Write set value].

Click [ Yes ].

- 67 -

Progress bar is displayed while writing.

Message addressing writing
complete is displayed.
Click [OK].

Set value of device parameter file is set to the connecting device.
Writing completed set value change its font color blue → black.

- 68 -

5-4. Operation of Acquisition
Acquisition provides functions of acquisition, recording (saves a file), displays real time trend and displays
real time data etc. for the data value which of device registered at 5-2-2. Device Registration Window andsupports acquisition.

< Name of acquisition parts at application launcher (data not yet acquired) >Utilizing following functions (1) Data Registration, (2) Group Registration and (3) Start Acquisition,
it processes data registration for acquisition, editing of group and start acquisition.
*Disabled tool button becomes enabled when acquisition starts.

(1)

(2)

(3)

No.

Name

Description

Reference(1)

Data Registration

Displays data acquisition window. It provides function of registration,

5-4-1

editing and deleting of data value for acquisition.
(2)

Group Registration

Displays group registration window. It provides editing function for

5-4-2

group for acquisition.
*Data for acquisition registered at (1) Data Registration is excluded
from data acquisition if not register to the group.
(3)

Start Acquisition

Starts data acquisition and periodical communication.

-< Name of acquisition parts at application launcher (during data acquisition) >Utilizing following functions (1) Stop Acquisition, (2) Acquisition Group Management, (3) Display List of
Alarm, (4) Data Memory Control, (5) Display Trend and (6) Display List of Data, it processes stopping
acquisition and monitoring acquisition etc.
*During acquisition, data registration, group registration and port registration and device registration of homefunction tool buttons become disabled.

(1)

(2)

(3)

(4)

(5)

(6)

No.

Name

Description(1)

Stop Acquisition

Stops data acquisition and periodical communication.

(2)

Acquisition GroupIt provides functions of recording (data memory) start, stop and control(3)

Display List of Alarm

Management

Reference
5-4-3

of acquisition group and monitoring of alarm status.
It provides functions of alarm status of acquisition group and each

5-4-4acquisition data, real time display and confirmation of alarm contents.
(4)

Data Memory

Clicking the drop down button displays list of acquisition group.

Control

Select and click a group desired to data memory control from thelist of acquisition group.

- 69 -

5-4-5

Click desired group to perform
data memory control.
Displays data memory control window.
(5)

Display Trend

Clicking the drop down button displays list of acquisition group.

5-4-6

Select and click a group desired to display trend graph from the listof acquisition group.

Click desired group to display
trend graph.
Displays trend graph display window.
(6)

Display List of Data

Clicking the drop down button displays list of acquisition group.
Select and click a group desired to display a list of data from the listof acquisition group.

Click desired group to
display list of data.
Displays a list of real time data display window.

- 70 -

5-4-7

5-4-1. Data Registration Window
It provides function of registration, editing and deleting for data value desired to acquire.
*It is for the data value which of device registered at 6-2-2. Device Registration Window and supports acquisition.

< Name of data registration window parts >Utilizing following functions (1) Edit (2) Copy, (3) Cut, (4) Paste, (5) Delete, (6) Data for Acquisition Setting
Information Display, (7) OK and (8) Cancel, it sets data for acquisition.

(1)

(2)

(3)

(4)

(5)

(6)

(7)

(8)

No.

Name

Description

Reference

(1)

Edit

Displays data edit dialog.

5-4-1-1(2)

Copy

Copies setting information of select area in the list of data for acquisition setting-

information.
*Shortcut is available with Ctrl + C key.
*Belonging group is initialized.
(3)

Cut

Cuts setting information of select area in the list of data for acquisition setting-

information.
*Shortcut is available with Ctrl +X key.
*Belonging group is initialized.
(4)

Paste

Pastes holding device information obtained by (2) Copy or (3) Cut after the-selected row.
*Shortcut is available with Ctrl +V key.
(5)

DeleteFrom registered data for acquisition setting information, deletes selecting data.

-*Deleting multiples is available.
(6)

Data for

Displays list of currently registered data for acquisition setting information.

Acquisition*Double clicking a cell enables setting of target row data for acquisition. Same as

Setting(1) Edit, it displays data edit dialog.

-

Information
Display(7)

OK

After starting up data registration window, reflects contents of registration, edit and-delete.
(8)

CancelAfter starting up data registration window, cancels contents of registration, edit anddelete.

- 71 --

5-4-1-1.Data Edit Dialog

It provides editing function of data registration window; data for acquisition setting information.

< Name of data edit dialog parts >Utilizing following functions (1) Device, (2) Data, (3) Data Name, (4) Unit, (5) Format, (6) Alarm 1 to 4,(7) Alarm Value of Alarm 1 to 4, (8) OK and (9) Cancel, it edits or cancels for data for acquisition setting
information.

(1)

(2)

(3)

(4)

(6)

(5)

(7)

(6)

(7)
(8)

No.
(1)

Name
Device(9)

Description
Select a device to perform data acquisition.
*Device registered at the section 6-2-2. Device Registration Window which has data for
acquisition can only be selected.

(2)

Data

Select an acquisition data type of device specified at (1) Device.
*Acquisition data type differs depending on the specified device.

(3)

Data Name

Specify acquisition data name for display. Up to 30 characters are able to be entered.

(4)

Unit

Specify acquisition data unit for display. Up to 30 characters are able to be entered.

(5)

FormatSelect and specify format for acquisition data display from #####, ####.#, ###.##,##.###, #.####, 0E+00, 0.0E+00, 0.00E+00 or 0.000E+00.

(6)

Alarm 1 to 4

Select and specify alarm type.
*Alarm type differs depending on the specified device and data for acquisition.

(7)

Alarm value ofSpecify alarm value for alarm specified at (6) Alarm 1 to 4 in the range of -99999.9999 to

Alarm 1 to 4

99999.9999.
*Alarm type: it is only enabled when computer judgment (upper limit) or computer
judgment (lower limit) is specified. Other alarms are depended on the devices.

(8)

OK

Reflects edited data for acquisition setting information on the data registration window.

(9)

CancelCancels edited data for acquisition setting information and returns to the prior to editing.

- 72 -

5-4-2. Group Registration Window
It provides functions of data adding, editing and deleting to the acquisition group.

< Name of group registration window parts >Utilizing following functions (1) Edit, (2) Group Setting Information Display, (3) OK and (4) Cancel,
it sets for acquisition group setting.

(1)
(2)

(3)

(4)

No.

Name

Description

Reference

(1)

Edit

Displays group edit dialog from selected acquisition group setting

5-4-2-1

information.
(2)

Group Setting

Displays list of acquisition group setting information.

Information*Double clicking a cell enables setting of target row acquisition group

Displaysetting. Same as (1) Edit, it displays group edit dialog.

(3)

OK

Reflects edited contents from startup of group registration window.

-(4)

CancelCancels edited contents from startup of group registration window.

-- 73 --

5-4-2-1. Group Edit Dialog

It provides functions of editing group registration window; acquisition group setting information.

< Name of group edit dialog parts >Utilizing following functions (1) Name, (2) Sample Count for Trend Display, (3) Interval for Trend Display,(4) Alarm Sound, (5) List of Registered Data, (6) OK and (7) Cancel, it edits or cancels for acquisitiongroup setting information.

(1)
(2)

(5)

(3)
(4)

No.

(6)

Name

(7)

Description

(1)

Name

Specify acquisition group name for display. Up to 30 characters are able to be entered.

(2)

Sample Count forSpecify trend buffer sample number at the acquisition in the range of 1 to 21600.

Trend Display(3)

Interval for Trend

Specify trend buffer interval at the acquisition in the range of 1 to 10.

Display(4)

Alarm Sound

Select and specify alarm sound from *.wav format file.

(5)

List of Registered

List of data for acquisition registered at the section 5-4-1. Data Registration Window isDatadisplayed.
*With check mark: Include in the group. Without check mark: exclude from the group.
*Multiple devices are not able to include in the group simultaneously.

(6)

OK

Reflects edited data for acquisition group setting information on the group registration
window.

(7)

CancelCancels edited data for acquisition group setting information and returns to the prior to
editing.

- 74 -

5-4-3. Acquisition Group Management Window
It provides functions of alarm status display of acquisition group and data memory start/stop.

< Name of acquisition group management window parts >Utilizing following functions (1) Data Memory Start, (2) Data Memory Stop and (3) List of Acquisition Group
Display, it manages and controls data group.

(2)

(1)

(3)

No.
(1)

Name

Description

Data Memory

Start data memory of selected row acquisition group.

Start*Enable/disable of the button switches at start/stop.
*Refer to 5-4-5 Data Memory Operation Window for more details.

(2)
(3)

Data Memory

Stop data memory of selected row acquisition group.

Stop*Enable/disable of the button switches at start/stop.

List of

Displays currently acquiring list of acquisition group information.

Acquisition

Information of group name, alarm status and memory status etc. is displayed.

Group Display[Alarm Status]
The icon below is displayed if there is an alarm activated data in the acquisition group.
…Alarm activation[Memory Status]
Four types of icon below describe data memory status.
…Stop (Data memory have not executed)…Recording (Data memory is executing and file is being outputted)…Waiting (Data memory is executing but file output is on the wait)*It only occurs when memory mode is on specify time.
…Abnormal (Data memory is failed for some reasons)*Error information is displayed on 5-2-3 Summary Display Window.

- 75 -

5-4-4. Alarm Display Window
It provides functions of alarm status display confirmation for each data for acquisition.

< Name of alarm display window parts >Utilizing following function (1) List of Alarm Display, it enables browsing and confirmation of list of alarm
display.

No.

Name

Description(1)

List of Alarm

Displays list of alarm activation and activated alarm. Activated alarm information of date,

Display

alarm, alarm type, belonging group, data name and acquisition data is displayed.
[Alarm]Three types of icon below describe alarm activation status.
…Alarm activation, unconfirmed by operator (red)…Alarm activation, confirmed by operator (blue)…Alarm activation → alarm unconfirmed by operator (yellow)*By clicking on the alarm row icon, it is judged that confirmation is done. At confirmation,change of icon or delete of alarm information is performed.

- 76 -

5-4-5. Data Memory Operation Window
It provides functions of each acquisition group data memory setting and control.

< Name of data memory operation window parts >Utilizing following functions (1) Acquisition Group Name, (2) Data Memory Start, (3) Data Memory Stop,(4) Data Memory Status, (5) Saving Folder Name, (6) Saving File Name, (7) Data Memory Interval,(8) Memory Mode Setting, (9) File Divide Setting and (10) Suffix, it performs data memory setting andcontrol.

(1)
(2)

(4)

(3)
(5)
(6)
(7)
(8)

(9)
(10)

No.

Name

Description(1)

Acquisition Group Name

Displays selected acquisition group name.

(2)

Data Memory Start

Start data memory.
*Enable/disable of the button switches at start/stop.

(3)

Data Memory Stop

Stop data memory.
*Enable/disable of the button switches at start/stop.

(4)

Data Memory Status

Displays data memory status of target acquisition group.
*According to the changes of status, display also changes in real-time.

(5)

Saving Folder Name

Specify saving folder name.
*Direct input is not allow.

(6)

Saving File Name

Specify saving file name.

- 77 -

(7)Data Memory Interval

Select and specify data memory interval from1, 5, 10 [second].

(8)

Memory Mode Setting

Select and specify memory mode from Manual or Specify Time.
*If select Specify Time, specify Starting Time and End Time.
*Same time cannot be specified for Starting time and Ending time. Also, set thetime as Starting time 00:00:00 and Ending time 23:59:59; only 1 sec. shift enables
continuous data memory.

(9)

File Divide Setting

Select and specify types of file divide from None, At starting, Every Month andEvery Day.
According to the specified condition, it performs file divide at data memory.
*This setting specify file divide (not divide) timing with suffix at data memory filesaving.
*If select None, set (10) Suffix.
*If select Every Month, specify date and time of file divide.
*If select Every Day, specify starting time of file divide.

(10)

Suffix

Select and specify suffix at file divide from Sequential number, yy/MM/dd/hh/mm/ss,
yy/MM/dd/hh/mm, yy/MM/dd/hh, yy/MM/dd or yy/MM.
*According to the specified suffix, the suffix is added to the saving file name.

- 78 -

5-4-6. Trend Graph Display Window
It provides functions of illustrating each acquisition group acquiring status graphically and display.

< Name of trend graph display window parts >Utilizing following functions (1) Acquisition Group Name, (2) Update Resume, (3) Pause Update,(4) Data Memory Start, (5) Data Memory Stop, (6) Data Setting, (7) Graph Area Setting,(8) Scale Plate Setting, (9) Numeric Data Setting, (10) Print, (11) Numeric Value Data Display,(12) Graph Area, (13) Scale Plate and (14) Time Axis Scale, it displays each acquisition group acquiring
status as a graph.

(2)

(3)

(4)

(1)
(5) (6)

(7)

(8)

(9)

(10)

(11)

(12)

(13)

(14)

No.

Name

Description

Reference(1)

Acquisition

Displays selected acquisition group name.

-

When update of trend graph is at pause, resume update.

-Group Name
(2)

Update Resume(3)

Update Pause*Enable/disable of the button switches at Updata/Pause.
During updating trend graph, pause data update.

-*Enable/disable of the button switches at Updata/Pause.
(4)

Data Memory

Start data memory.

Start*Enable/disable of the button switches at Start/Stop.

Data Memory

Stop data memory.

Stop*Enable/disable of the button switches at Start/Stop.

(6)

Data Setting

Displays Data Setting dialog.

(7)

Graph Area

Displays Graph Area Setting dialog.

Setting*Setting information differs depending on the each group.

Scale Plate

Displays Scale Plate Setting dialog.

Setting*Setting information differs depending on the each group.

(5)

5-4-6-1*Setting information differs depending on the each group.

(8)- 79 -

5-4-6-2
5-4-6-3(9)(10)

Numeric Value

Displays Numeric Value Data Setting dialog.

Data Setting*Setting information differs depending on the each group.

Print

Prints screen image.

5-4-6-4-*Displays Print Preview.
(11)

Numeric Value

Displays information of data number, legend, data name, numeric

Data Display

value data, unit, scale number and alarm status etc. for each acquiring

-

data.
(12)

Graph Area

Illustrates acquiring data in real-time.

-

(13)

Scale Plate

Displays scale which indicates temperature area of trend graph.

-*Specify the range by dragging with mouse magnifies the area.
(14)

Time Axis Scale

Displays scale which indicates time axis of trend graph. *Specify therange by dragging with mouse magnifies the area.

- 80 --

5-4-6-1. Data Setting Dialog

It provides editing function of trend graph display window; data setting information.

< Name of data setting dialog parts >Utilizing following functions (1) Data No., (2) Data Name, (3) Display, (4) Scale Plate, (5) Dot Setting, (6)Line Setting (7) OK and (8) Cancel, it edits or cancels for trend graph display setting information.

(1)

(2)

(3)

(4)

(6)

(5)

(7)
No.

Name

(8)

Description(1)

Data No.

Displays data No. It cannot be changed.

(2)

Data Name

Displays data name. It cannot be changed.

(3)

Display

Specify display or hide display.
*With check mark: Displays/Without check mark: Hidden(4)

Scale Plate

Select and specify target scale plate from Scale 1, Scale 2, Scale 3 or Scale 4 for the data
value to plot.

(5)

Dot SettingSpecify types, colors or size of the dot to displays.
[Type]
Select and specify from None,

,

,

,

,

,

,

, ,

,

or

.

[Color]Specify from color dialog.
[Size]
Select and specify from Small, Normal or Large.
(6)

Line SettingSpecify types, colors or width of the line to display.
[Type]
Select and specify from None, Solid line, Chain dash line, Wave line, Dash dotted line or
Dash double dotted line.
[Color]Specify from color dialog.
[Width]Select and specify from Thin, Normal or Thick.

(7)

OK

Retains setting information on the data setting dialog and reflects setting contents on thetrend graph display.

(8)

CancelIgnores setting information on the data setting dialog and cancels setting contents on thetrend graph display.

- 81 -

5-4-6-2. Graph Area Setting Dialog

It provides editing function of trend graph display window; graph area setting information.

< Name of graph area setting dialog parts >Utilizing following functions (1) Drawing Directions, (2) Scroll Mode, (3) Display Border, (4) Reference
Scale,(5) Border Color, (6) Font, (7) Background Color, (8) Time Color, (9) OK and, (10) Cancel ,it edits or cancels for trend graph; graph area setting information.

(1)

(2)

(3)

(6)

(4)

(5)

(8)

(7)
(9)

No.
(1)

Name
Drawing(10)

Description
Select and specify scroll method of the graph from Horizontal or Vertical.

Directions(2)

Scroll Mode

Select and specify scroll method from Auto or Specify Time.
*If select Specify Time, select and specify displaying time interval from 1 to 59 minutes,1 to
23 hours, 1 to 31 days or 1 to 12 months.

(3)

Display Border

Select and specify displaying border type from None, Normal (drawn as scale) or Detail(drawn also as sub scale).

(4)

Reference

Select and specify target scale plate for displaying border from 1, 2, 3 or 4.

Scale
(5)

Border Color

Specify the color of drawing border.

(6)

Font

Specify the font used in graph area and time axis.

(7)

Background

Specify the background color of graph area.

Color
(8)

Time Color

Specify the foreground color of time axis.

(9)

OK

Retains setting information on the graph area setting dialog and reflects setting contents on
the trend graph display window.

(10)

CancelIgnores setting information on the graph area setting dialog and cancels reflecting setting
contents on the trend graph display window.

- 82 -

5-4-6-3. Scale Plate Setting Dialog

It provides editing function of trend graph display window; scale plate setting information.

< Name of scale plate setting dialog parts >Utilizing following functions (1) Scale Plate, (2) Display, (3) Display Position, (4) Logarithm, (5) Scale
Upper Limit, (6) Scale Lower Limit, (7) Unit, (8) Scale Dividing Number, (9) Background Color, (10) Scale
Color, (11) Font Color, (12) Font, (13) OK and (14) Cancel, it edits or cancels for scale plate setting
information used on trend graph.

(1) (2) (3) (4) (5)

(6)

(7)

(8)

(9) (10) (11) (12)(13) (14)
No.

Name

Description(1)

Scale Plate

Displays 1 to 4 scale plate No. It cannot be changed.

(2)

Display

Specify display or hide display.
*With check mark: Displays/Without check mark: Hidden(3)

Display Position

Select and Specify scale plate display position from Right (horizontal)/Down (vertical).
*By changing Drawing Direction on 5-4-6-2. Graph Area Setting Dialog, Left/Up or
Right/Down switches automatically.

(4)

Logarithm

Select and specify scale value expression from normal or Logarithm.
*With check mark: Logarithm/Without check mark: normal(5)

Scale Upper

Specify scale plate upper limit in the range of -99999.999 to99999.999.

Limit*For logarithm scale, it becomes exponential format.

Scale Lower

Specify scale plate lower limit in the range of -99999.999 to99999.999.

Limit*For logarithm scale, it becomes exponential format.

(7)

Unit

Specify unit to display on scale plate. Up to 30 characters are able to be entered.

(8)

Scale Dividing

Select and specify number of scale line division from 2, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14,

Number15, 16, 17, 18, 19 and 20.

(6)*9 are fixed for logarithm scale.
(9)

Background

Specify the background color of scale plate.

Color
(10)

Scale Color

Specify the scale line drawing color.

(11)

Font Color

Specify the scale value foreground color.

(12)

Font

Specify the scale value font.

(13)

OK

Retains setting information on the scale plate setting dialog and reflects setting contents on(14)

Cancelthe trend graph display window.
Ignores setting information on the scale plate setting dialog and cancels reflecting setting
contents on the trend graph display window.

- 83 -

5-4-6-4. Numeric Data Setting Dialog

It provides editing function of trend graph display windo; scale plate setting information.

< Name of numeric data setting dialog parts >Utilizing following functions (1) Display Numeric Data, (2) Display Item, (3) Font, (4) Font Color, (5)Background Color, (6) Display Position, (7) OK and (8) Cancel, it edits or cancels for numeric data setting
information.

(1)
(2)

(3)
(4)

(5)
(6)

(7)

(8)

No.

Name

(1)

Display NumericSpecify display or hide numeric data.

Data*With check mark: Displays/Without check mark: Hidden

Display Item

Specify items for displaying of numeric data.

(2)

Description*With check mark: Displays/Without check mark: Hidden
(3)

Font

Specify font information of numeric data.

(4)

Font Color

Specify foreground color of numeric data.

(5)

Background

Specify background color of numeric data.

Color
(6)

Display Position

Select and Specify numeric data display position from Top, Bottom, Right or Left.

(7)

OK

Retains setting information on the numeric data setting dialog and reflects setting contents(8)

Cancelon the trend graph display window.
Ignores setting information on the numeric data setting dialog and cancels setting contentson the trend graph display window.

- 84 -

5-4-7. List of Real Time Data Display Window
It provides functions of expand list of acquiring data included in the selected acquiring group then update
and display in real-time.

<Name of list of real time data display window parts>
Utilizing following functions (1) Acquisition Data Name, (2) Update Resume, (3) Update Pause, (4) Magnify,(5) Reduce, (6) Display Setting, (7) Print and (8) List of Real Time Data Display, it displays and updates listof acquiring data.

(1)(2) (3) (4) (5) (6) (7)

(8)

No.

Name

Description

Reference(1)

Acquisition Data

Displays selected acquisition group display name.

-

When update of list of data is at pause, resume update.

-

Name
(2)

Update Resume*Enable/disable of the button switches at Pause/Update.
(3)

Update Pause(4)

MagnifyDuring updating the list of data, pause data update.

-*Enable/disable of the button switches at Pause/Update.
Magnifies font size of displayed list.

-*Magnify twice as much as standard value in maximum.
(5)

Reduce(6)

Display Setting

Reduces font size of displayed list.

-*Reduce 0.5 times of standard value in maximum.
Displays display setting of list of real time data.

5-4-6-1*It can be displayed by double clicking on the (8) List of RealTime Data Display.
(7)

Print

Prints displaying list of data.

-*Displays Print Preview.
(8)

List of Real Time Data

Displays list of acquiring data included in the selected acquiring

Displaygroup in real-time.

- 85 --

5-4-7-1. Display Setting Dialog

It provides editing function of list of real time data display window; display setting information.

< Name of display setting dialog parts >Utilizing following functions (1) Display Item, (2) Font, (3) Odd Number Row, Cell Setting, (4) Even Number
Row, Cell Setting, (5) Cell Setting at Alarm Activation, (6) OK and (7) Cancel, it edits or cancels for list ofreal time data; display setting information.

(1)

(2)
(3)
(4)
(5)
(6)

No.

Name

(7)

Description(1)

Display Item

Specify display or hide for each column of list of real time data.

(2)

Font

Specify the font information of list of real time data display window.

(3)

Odd Number Row,

Specify the font color and the background color of odd number row cell of list of real

Cell Setting

time data display window.

Even Number Row,

Specify the font color and the background color of even number row cell of list of real

Cell Setting

time data display window.

Cell Setting at Alarm

Specify the font color and the background color of alarm activation cell of list of real

Activation

time display window.

OK

Retains setting information on the display setting dialog and reflects setting contents*With check mark: Displays/Without check mark: Hidden

(4)
(5)
(6)

on the list of real time data display window.
(7)

CancelIgnores setting information on the display setting dialog and cancels reflecting setting
contents on the list of real time data display window.

- 86 -

5-5. Operation of Data Analysis
Data analysis provides function of expand, display and analyze recorded acquisition data file in thisapplication, acquired data file recorded by each device side and analysis file saved at this function.

< Name of data analysis at application launcher >Utilizing following (1) Data Analysis, it processes expand, display and analysis of acquisition data file etc.

(1)

No.

Name

(1)

Data
Analysis

Description

Reference

Displays file open dialog to search acquisition data file etc.

5-5-1Displays data analysis window by using specified data file.
If combining acquisition data files is desired, select multiple files whiledisplaying file open dialog.

*Combining acquisition data file requires following conditions (Refer to 5-4-2-1. Group Edit Dialog and 5-45. Data Memory Operation Window for condition setting).
 Combining acquisition data files shall be same group configurations (group name, data configuration anddata memory interval shall be the same).
 Combining acquisition data files shall be same extensions.
*If the selected acquisition data files are not fulfilling the combining conditions, combining is not carried out
and files are displayed on the other windows.

- 87 -

5-5-1. Data Analysis Window
It provides function of expand, display and analyze acquired data by this application, each device side or
analysis file saved at analysis function.

< Name of data analysis window parts >Utilizing following functions (1) Data Analysis Window Title, (2) Toolbar (3) Numeric Data/Bar Graph
Display and (4) Trend Graph Display, it expands, displays and analyzes target analysis file.

(1)

(2)

(3)

(4)

No.

Name

Description

Reference(1)

Data Analysis

Displays in order of Analysis – [analysis file title property of analysis

Window Titlefile] – name of analysis file.

-*Default of analysis data file name is group name of currently open
data.
(2)

Toolbar

Execute each function of data analysis window.

5-5-1-1(3)

Numeric Data/Bar

Displays data which links to cursor etc. on expended and displayed

5-5-1-2Graph Display(4)

Trend Graphtrend graph.
Displays acquisition data value of analysis data file as a graph.

Display- 88 -

5-5-1-3

5-5-1-1. Toolbar

It provides function of tool buttons which startup and execute each function of data analysis window and its
switch function.

< Each toolbar, operation flow >(1) Startup data analysis window.
When startup and display data analysis window, it displays [Home] menu.

(2) Select a function.

Broadly three functions are available to select.
From left of menubar, [Home], [Display] and [Setting] are switched to relevant tool buttons by clicking on amenu.
Click on the [Display].

A list of display function tool buttons is displayed.

- 89 -

(3) Starting up each function.

After selecting a function at (2), click a tool button in below.
Clicking on a tool button enables startup relevant function and display a dialog/window.
Click on a display list of data button.

Displays clicked tool button relevant dialog
or executes a function.

- 90 -

(a) Operation of Home

It provides basic function of data analysis.

< Name of home toolbar parts >Utilizing following functions (1) Open File, (2) Save, (3) Save as, (4) Display List of Data, (5) Print Screen
Image, it performs basic operation of data analysis.

(1)

(2)

No.

Name

(1)

Open File

(3)

(4)

(5)

Description

Reference

Displays file open dialog.

-

Specify data file, expands and displays data analysis
window.
*1 *.dmf file or *.zil file can be selected.
*2 Data values which are able to combine iscombined to original data then expanded anddisplayed. For other case, another window startup
then expanded and displayed.
*3 If the number of the windows exceeds prescribed
displayable number, it cannot be startup.
(2)

Save

Saves analysis data file.

-*The file was saved as zil format, it saves as sameformat and in other case, it saves the file with a new
name.
(3)

Save asSpecify arbitrary folder and saves the file with a new

-

name.
(4)

Display List of Data

Displays list of data dialog.

(a)-1.

(5)

Print Screen Image

Prints current screen image.

-*Displays print preview.

- 91 -

(a)-1. Data List Display Window
It provide function of expand and display for data value of data analysis window trend graph as a list ofnumeric value.
< Name of data list display window parts >Utilizing following functions (1) Export, (2) Display Setting, (3) Print and (4) Data List Display, it displays listof data analysis.

(1) (2) (3)
(4)

No.

Name

Description(1)

ExportSelect a file format from

Reference-

CSV (Comma delimited)
Text (Tab delimited)
Excel (xlsx)
and specify output destination then output a file.
*A graph is added only for Excel (xlsx) format.
(2)

Display

Displays display setting dialog.

(a)-1-1.

Setting(3)

Print

Prints displayed data list.

-*Displays print preview.
(4)

Data List
Display

Displays list of acquisition time and a list of acquisition
value.
*Double click on the data list display displays a cursor on
the 5-5-1-3. Trend Graph Display. Refer to the 5-5-1-3.
Trend Graph Display (2) List Cursor for more detail.

- 92 --

(a)-1-1. List Display Setting Dialog
It provides editing function of data list display; display setting information.
< Name of list display setting dialog parts >Utilizing following functions (1) Display Interval Setting, (2) Font, (3) Odd Number Row Setting, (4) Even
Number Row Setting, (5) OK, and (6) Cancel, it edits for display setting information.

(1)

(2)
(3)

(4)
(6)

(5)
No.
(1)

Name

Description

Display IntervalSpecify usage of display interval and setting for display interval of time for data value toSettinglist display from the range of 1 to 86400 [sec].
*Display data list by the specified interval on the data list display window according to thedisplay interval.

(2)

Font

Specify the font information of data list display window.

(3)

Odd Number Row

Specify the font color and background color of odd number row on the data list display

Setting

window.

Even Number

Specify the font color and background color of even number row on the data list display

Row Setting

window.

OK

Retains setting information on the list display setting dialog and reflects setting contents

(4)
(5)

on the data list display window.
(6)

CancelIgnores setting information on the list display setting dialog and cancels reflecting setting
contents on the data list display window.

- 93 -

(b) Operation of Display

It provides function of switching display setting in data analysis.

< Name of display toolbar parts >Utilizing following functions (1) Display Period Switching, (2) Border Display Switching, it switches display
setting of data analysis.

(1)

(2)

No.

Name

(1)

Display Period
Switching

Description
Select and specify display period on the trend graph from All period, Standard or
Between AB cursor.
Click on the dropdown button then specify one of the icons below by clicking.
*While selecting Between AB cursor, clicking on the icon part of its button changes
display area (on the display, place the AB cursor → click icon).
…All Period…Standard…Between AB cursor(2)

Border Display
SwitchingSelect and specify border display on the trend graph from None, Standard or Detail.
Click on the dropdown button then specify one of the icons below by clicking.
…None…Standard…Detail- 94 -

(c) Operation of Setting

It provides function of various setting for data analysis.

< Name of setting toolbar parts >Utilizing following functions (1) Analysis File Setting, (2) Data Setting, (3) Graph Area Setting, (4) Scale
Plate Setting, and (5) Numeric Data/Bar Graph Setting, it edits setting information of data analysis.
*Refer to the section 3-1 List of function restriction of free of charge edition about restriction of free of charge
edition.

(1)

(2)

(3)

(4)

(5)

No.

Name

Description

Reference(1)

Analysis File Setting

Displays analysis file setting dialog.

(c)-1.

(2)

Data Setting

Displays data setting dialog.

(c)-2.

(3)

Graph Area Setting

Displays graph area setting dialog.

(c)-3.

(4)

Scale Plate Setting

Displays scale plate setting dialog.

(c)-4.

(5)

Numeric Data/Bar

Displays numeric data/bar graph setting dialog.

(c)-5.

Graph Setting- 95 -

(c)-1. Analysis File Setting Dialog
It provides editing function of data analysis window, analysis file setting information.
< Name of analysis file setting dialog parts >Utilizing following functions (1) Analysis File Title, (2) Data Number, (3) Field Number, (4) Data Name, (5)Unit, (6) Format, (7) Alarm 1 to 4 Name, (8) OK, and (9) Cancel, it edits analysis file setting information.

(1)
(3)

(2)

(4)

(5)

(6)

(7)

(8)

No.

Name

(1)

Analysis File Title(9)

Description
Specify analysis file title which is expanded on the data analysis window.
Up to 30 characters are able to be entered.
*Specified analysis file title is reflected on the data analysis window header text
string.

(2)

Data Number

Displays data number which exists on the data analysis window.
It cannot be changed.

(3)

Field Number

Displays data No.at acquisition. It cannot be changed.

(4)

Data Name

Specify data name. Up to 30 characters are able to be entered.

(5)

Unit

Specify unit. Up to 30 characters are able to be entered.

(6)

FormatSelect and specify display format for analysis data from #####, ####.#, ###.##,##.###, #.####, 0E+00, 0.0E+00, 0.00E+00 or 0.000E+00.

(7)

Alarm 1 to 4 Name

Specify each alarm name. Up to 30 characters are able to be entered.

(8)

OK

Retains setting information on the analysis file setting dialog and reflects setting(9)

Cancelcontents on the data analysis window.
Ignores setting information on the analysis file setting dialog and cancels reflecting
setting contents on the data analysis window.

- 96 -

(c)-2. Data Setting Dialog
It provides editing function of data analysis window; data setting information.
< Name of data setting dialog parts >Utilizing following functions (1) Basic Setting, (2) Alarm Setting, (3) OK and (4) Cancel, it edits or cancels
data analysis window; data setting information.

(1)

(2)

(3) (4)No.

Name

Description(1)

Basic Setting*Refer to name of basic setting parts.

(2)

Alarm Setting*Refer to name of alarm setting parts.

(3)

OK

Retains setting information on the data setting dialog and reflects setting contents on thedata analysis window.

(4)

CancelIgnores setting information on the data setting dialog and cancels reflecting setting contentson the data analysis window.

< Name of basic setting parts >Utilizing following functions (1) Data No., (2) Data Name, (3) Display (4) Scale Plate, (5) Dot Setting and(6) Line Setting, it edits data analysis window; data setting information; basic setting.

(2)

(1)

No.

(3)

(4)

(6)

(5)

Name

Description(1)

Data No.

Displays data No.at acquisition. It cannot be changed.

(2)

Data Name

Displays data name. It cannot be changed.

(3)

Display

Specify display or hide display data value.
*With check mark: Displays/Without check mark: Hidden(4)

Scale Plate(5)

Dot Setting

Select and specify target scale plate for data value to plot from Scale 1, Scale 2, Scale 3 or
Scale 4.
Specify type, color and size of displaying dot.
[Type]
Select and specify from None/

/

/

/

/

[Color]Specify from color dialog.
[Size]
Select and specify from Small, Normal or Large.

- 97 -

/

/

/

/

/

/

(6)

Line SettingSpecify type, color and width of displaying line.
[Type]
Select and specify from None, Solid line, Chain dash line, Wave line, Dash dotted line or
Dash double dotted line.
[Color]Specify from color dialog.
[Width]Select and specify from Thin, Normal or Thick.

< Name of alarm setting parts >Utilizing following function (1) Alarm 1 to 4 Setting, it edits data analysis window; data setting information;
alarm setting.

(1)

No.

Name

(1)

Alarm 1 to 4 Setting

Description
Specify type, color, width and its priority of displaying dot for each alarm.
[Priority]
Select and specify from 1, 2, 3 or 4.
*Priority is 1 [high] to 4 [low].
[Type]
Select and specify from None/

/

/

/

/

[Color]Specify from color dialog.
[Size]
Select and specify from Small, Normal or Large.

- 98 -

/

/

/

/

/

/

(c)-3. Graph Area Setting Dialog
It provides editing function of data analysis window; graph area setting information.
< Name of graph area setting dialog parts >Utilizing following functions (1) Drawing Direction, (2) Font, (3) Background Color, (4) Color of Time,(5) Display Border, (6) Reference Scale, (7) Border Color, (8) Cursor A Setting, (9) Cursor B Setting,(10) List Cursor Setting, (11) OK, and (12) Cancel, it edits or cancels data analysis window; graph area
setting information.

(8)

(1)
(2)
(3)

(4)

(5)

(6)

(9)
(10)

(7)
(11)
No.
(1)

Name
Drawing(12)

Description
Select and specify scroll method of the graph from Horizontal or Vertical.

Direction
(2)

Font

Specify the font using at graph area and time axis.

(3)

Background

Specify the background color of graph area.

Color
(4)

Color of Time

Specify the foreground color of time axis.

(5)

Display Border

Select and Specify display type of border from None, Normal (drawn as scale) or Detail(drawn also as sub scale).

(6)Reference Scale

Select and specify target scale plate for displaying border from 1, 2, 3 or 4.

(7)

Border Color

Specify the color of drawing border line..

(8)

Cursor A SettingSpecify line width and line color of cursor A.
[Line Width]
Select and specify from Thin, Normal or Thick.
[Line Color]
Specify from color dialog.

(9)

Cursor B SettingSpecify line width and line color of cursor B.
[Line Width]
Select and specify from Thin, Normal or Thick.
[Line Color]
Specify from color dialog.

- 99 -

(10)List Cursor

Specify line width and line color of list cursor.

Setting[Line Width]
Select and specify from Thin, Normal or Thick.
[Line Color]
Specify from color dialog.

(11)

OK

Retains setting information on the graph area setting dialog and reflects setting contents on
the data analysis window.

(12)

CancelIgnores setting information on the graph area setting dialog and cancels reflecting setting
contents on the data analysis window.

- 100 -

(c)-4. Scale Plate Setting Dialog
It provides editing function of data analysis window; scale plate setting information.
< Name of scale plate setting dialog parts >Utilizing following functions (1) Scale Plate, (2) Display, (3) Display Position, (4) Logarithm, (5) Scale
Upper Limit, (6) Scale Lower Limit, (7) Unit, (8) Scale Dividing Number, (9) Background Color, (10) Scale
Color, (11) Font Color, (12) Font, (13) OK and (14) Cancel, it edits or cancels for scale plate setting
information used on trend graph.

(1) (2) (3) (4) (5)

(6)

(7)

(8)

(9) (10) (11) (12)(13) (14)
No.

Name

Description(1)

Scale Plate

Displays scale No. 1 to 4. It cannot be changed.

(2)

Display

Specify display or hide display of scale plate.
*With check mark: Displays/Without check mark: Hidden(3)

Display Position

Select and Specify scale plate display position from Right (horizontal)/Down (vertical).
*By changing Drawing Direction on 5-4-6-2. Graph Area Setting Dialog, Left/Top or
Right/Bottom switches automatically.

(4)

Logarithm

Select and specify scale value expression from normal or Logarithm.
*With check mark: Logarithm/Without check mark: normal.

(5)

Scale Upper

Specify scale plate upper limit in the range of -99999.999 to 99999.999.

Limit*For logarithm scale, it becomes exponential format.

Scale Lower

Specify scale plate lower limit in the range of -99999.999 to 99999.999.

Limit*For logarithm scale, it becomes exponential format.

(7)

Unit

Specify unit to display on scale plate. Up to 30 characters are able to be entered.

(8)

Scale Dividing

Select and specify number of scale line division from 2, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14,

Number15, 16, 17, 18, 19 and 20.

(6)*9 are fixed for logarithm scale.
(9)

Background

Specify the background color of scale plate.

Color
(10)

Scale Color

Specify the scale line drawing color.

(11)

Font Color

Specify the scale value foreground color.

(12)

Font

Specify the scale value font.

(13)

OK

Retains setting information on the scale plate setting dialog and reflects setting contents on(14)

Cancelthe trend graph display window.
Ignores setting information on the scale plate setting dialog and cancels reflecting setting
contents on the trend graph display window.

- 101 -

(c)-5. Numeric Value Data/Bar Graph Setting Dialog
It provides editing function of data analysis window; numeric value data/bar graph setting information.
< Name of numeric value data /bar graph setting dialog parts >Utilizing following functions (1) Display or Hide Numeric Value Data/Bar Graph, (2) Display Position,(3) Types of Legend, (4) Numeric Value Data, Display Item, (5) Numeric Value Data, Font, (6) Numeric
Value Data, Font Color, (7) Numeric Value Data, Background Color, (8) Numeric Value Data, All Data
Setting, (9) Numeric Value Data, Data between Cursor Setting (10) Bar Graph, Direction of Legend, (11)
OK and (12) Cancel, it edits or cancels for numeric value data/bar graph setting information.

(1)
(2)
(3)
(5)

(4)

(6)

(7)

(8)
(9)

(10)
(11)

(12)

No.

Name

Description(1)

Display or Hide

Specify display or hide numeric value data and bar graph.

Numeric Value*With check mark: Displays/Without check mark: Hidden

Data/Bar Graph(2)

Display Position

Select and specify display position of numeric data and bar graph from Top, Down, Left or
Right.

(3)

Types of Legend

Select and specify types of legend from Numeric value data, Bar graph or Stacked bar
graph.
*When numeric value data is selected, numeric value data setting becomes enabled andbar graph setting becomes disabled.
*When bar graph or stacked bar graph is selected, bar graph setting becomes enabled andnumeric value data setting becomes disabled.

- 102 -

(4)Numeric Value

Select and specify numeric value data display item.

Data, Display*With check mark: Displays/Without check mark: Hidden

Item
(5)

Numeric Value

Specify font information of numeric value data.

Data, Font
(6)

Numeric Value

Specify foregrround color of numeric value data.

Data, Font Color(7)

Numeric Value

Specify background color of numeric value data.

Data,
Background
Color
(8)

Numeric Value

Specify font color and background color of numeric value data, all data related strings.

Data, All Data

*Default background color is located at “Custom color” in the color selecting dialog.

Setting(9)

Numeric Value

Specify font color and background color of numeric value data, data between cursor

Data, Data

related strings.

between Cursor*Default background color is located at “Custom color” in the color selecting dialog.

Setting(10)

Bar Graph,Specify bar graph direction.

Direction ofLegend(11)

OK

Retains setting information on the numeric value data setting dialog and reflects setting
contents on the trend graph display window.

(12)

CancelIgnores setting information on the numeric value data setting dialog and cancels reflecting
setting contents on the trend graph display window.

- 103 -

5-5-1-2. Numeric Value Data/Bar Graph Display

It provides display area for numeric value data or bar graph which accumulated from expanded data on
data analysis window.
*It is interlocked with 5-5-1-3. Trend Graph Display therefore moving of AB cursors on the trend graph performs

recalculation and so changes display contents.

< Name of numeric value data parts >Utilizing following functions (1) Basic Data, Cursor AB Data Display and (2) All Data, Data between Cursor
Display, it displays data analysis window, numeric value data.

(1)
No.
(1)

(2)

Name

Description

Basic Data, Cursor AB*Refer to <Name of basic data, cursor AB data> below.

Data Display*Double click on the display displays (c)-5. Numeric Value Data/Bar Graph Setting
Dialog.

(2)

All Data, Data between*Refer to <Name of all data, data between cursor display> below.

Cursor Display*Double click on the display displays (c)-5. Numeric Value Data/Bar Graph Setting
Dialog.

< Name of basic data, cursor AB data parts >Utilizing following functions (1) Data Name, (2) Legend, (3) Data Number, (4) Cursor A, (5) Cursor B, (6)Cursor A to B, (7) Unit, (8) Scale, (9) Cursor A Alarm and (10) Cursor B Alarm, it configures numeric value
data display.

(1)

(2)

No.

(3)

(4)

(5)

(6)

Name

(7)

(8)

(9)

(10)

Description(1)

Data Name

Displays data name as row header.

(2)

LegendDisplays legend line corresponding to each data value.

(3)

Data Number

Displays data number at acquisition for each data value.

(4)

Cursor A

Displays value of cursor A position within each data value. For time row, time of target
data value is displayed. Displayed color next to the header shows color of the cursor.
*It is displayed at the same time when cursor AB is placed at 5-5-1-3. Trend Graph
Display. Data value is deleted at the same time when cursor A is deleted.

(5)

Cursor B

Displays value of cursor B position within each data value. For time row, time of target
data value is displayed. Displayed color next to the header shows color of the cursor.
*It is displayed at the same time when cursor AB is placed at 5-5-1-3. Trend Graph
Display. Data value is deleted at the same time when cursor B is deleted.

(6)

A-B

Displays difference which subtracts cursor B value from cursor A value while (4) and (5)are displayed.
*It is displayed at the same time when cursor AB is placed at 5-5-1-3. Trend Graph
Display. Calculation result is deleted at the same time when cursor A or cursor B isdeleted.

(7)

Unit

Displays unit specified for each data value.

(8)

Scale

Displays scale plate, scale information specified for each data value.

- 104 -

(9)Cursor A Alarm

Displays alarm status of cursor A position within each data value.

(10)

Cursor B Alarm

Displays alarm status of cursor B position within each data value.

< Name of all data, data between cursor display parts >Utilizing following functions (1) All Data Maximum Value, (2) All Data Minimum Value, (3) All Data, Average
Value, (4) All Data Standard Deviation, (5) All Data Median Value, (6) Cursor Maximum Value, (7) All Data
Minimum Value, (8) Cursor Average Value, (9) Cursor Standard Deviation and (10) Cursor Median Value, it
configures numeric value data display.

(2)

(1)

No.
(1)

(3)

(4)

(5)

Name
All Data Maximum

(6)

(7)

(8)

(9)

Description
Displays maximum value within all data values for each data value.

Value
(2)

All Data Minimum

Displays minimum value within all data values for each data value.

Value
(3)

All Data, Average

Displays average value within all data values for each data value.

Value
(4)

All Data Standard

Displays standard deviation value within all data values for each data value.

Deviation(5)

All Data Median

Displays median value within all data values for each data value.

Value
(6)

Cursor Maximum

Displays maximum value between cursor AB for each data value.

Value
(7)

All Data Minimum

Displays minimum value between cursor AB for each data value.

Value
(8)

Cursor Average

Displays average value between cursor AB for each data value.

Value
(9)

Cursor Standard

Displays standard deviation value between cursor AB for each data value.

Deviation(10)

Cursor Median

Displays median value between cursor AB for each data value.

Value- 105 -

(10)

5-5-1-3. Trend Graph Display

It provides functions of illustrating graphically and displaying for retained acquisition data value of data
analysis window, analysis data file.
*It is interlocked with 5-5-1-2. Numeric Value Data/Bar Graph Display therefore moving of AB cursors on the trend
graph performs recalculation and so changes display contents.

< Name of trend graph display parts >Utilizing following functions (1) Graph Area, (2) Cursor A, (3) Cursor B, (4) List Cursor, (5) Scale Plate Area
and, (6) Time Axis Area, it displays trend graph and analysis on the data analysis window.

(2)

(1)

(3)

(4)

(5)

Acquisition data value

(6)
No.

Name

Description

Reference(1)

Graph Area

Illustrates analysis data value as graph and displays graph-etc.
*Border line, A cursor and B cursor which conform reference
scale are displayed.
*Right clicking on the graph area displays Add message of(5) Message.
(2)

Cursor A

Clicking on the mouse on the (1) Graph Area displays A-

cursor.
*Cursor A being selected status enables moving the cursor
by mouse drag or press of arrow key and delete the cursor
by press of a Delete key.
(3)

Cursor B

Cursor B is displayed by mouse drag after displaying (3)-Cursor A*Cursor B being selected status enables moving the cursor
by mouse drag or press of arrow key and delete the cursor
by press of a Delete key.
(4)

List Cursor

Double click on the (a)-3. Data Display Window (5) Data List
Display displays a cursor on the corresponding data location.
*List cursor disappears when data list display is closed.

- 106 --

(5)Scale Plate Area

Displays border line reference scale of (1) Graph Area.
*Editing scale plate upper limit/lower limit etc. setting isavailable at (c)-4. Scale Plate Setting Dialog.
*If multiple scale plates are displayed, mouse click on thedesired scale plate to set border line reference scale.
*Attention is required for each data value reference scale
since it is the setting contents of (c)-2. Data Setting Dialog.
*Specify the range by dragging with mouse magnifies the
area.

(6)

Time Axis Area

Displays time axis of (1) Graph Area.
*Specify the range by dragging with mouse magnifies thearea.

- 107 --

5-6. Operation of Favorite
Favorite provides function of user support by adding arbitrary function which is enabled on launcher of thisapplication so the function with high frequency of use can startup and execute smoothly.

< Name of favorite parts at application launcher >Processing (1) Favorite Tool Button Display.

(1)

No.
(1)

Name
Favorite Tool Button Display

Description
Displays tool button which user added or registered arbitrary.
*Refer to <Flow of adding favorite>.

- 108 -

< Flow of adding favorite >Add arbitrary tool button in the menu of [Favorite] by following and operate figures below.

Right click on desired tool button to add to the favorite.

Submenu is displayed. Click Add to Favorite.

Click on the [Favorite] button.

Device Registration tool button is added to the menu of the [Favorite].

- 109 -

< Flow of creating shortcut >Create arbitrary tool button shortcut by following and operate figures below.
*Shortcut is able to be located on the desktop or specify a file name and put in the arbitrary place.

Right click on desired tool button to create the shortcut.

Submenu is displayed. Click [Create shortcut] → [Specify file name].

Specify shortcut registering destination and click Save button.

- 110 -

Click on the created shortcut startup the application and executetarget function.
*If the application is already running, it is invalid.

- 111 -

6. TroubleshootingThis section describes symptoms, cause and remedy when a trouble is found in operation or function ofthis software.
No.
1

Symptom
Installation does not complete

Cause and Remedy
Confirm that there is enough free hard disk space (refer to the section 2-1).

correctly.
2

The software does not startup.

The installation might not be completed correctly. Uninstall the software theninstall it again, and check if the software is able to startup.

3

The software has been operating Exit the software, then startup the software again.
correctly, but suddenly amalfunction occurs.

*If the software exits due to an unexpected error, unsaved data would not bebacked up. Please set it again.

- 112 -

CHINO CORPORATION
32-8, KUMANO-CHO, ITABASHI-KU, TOKYO 173-8632
Telephone: 81-3-3956-2171
Facsimile: 81-3-3956-0915
Printed in Japan