ScheduleIssue date: 19 July 2017
Valid until: 19 July 2020
MS ISO/IEC 17025

NO: SAMM 839

Page: 1 of 3

LABORATORY LOCATION:(PERMANENT LABORATORY)

Scan this QR Code or visit www.jsm.gov.my/cab-directories for the current scope of accreditation

FIELD OF CALIBRATION:LABORATORY OF NIHON DENKEI (M) SDN. BHD.
UNIT NO. A-11-1, NORTH POINT OFFICE
MID VALLEY CITY
NO. 1, MEDAN SYED PUTRA UTARA
59200 KUALA LUMPUR
ELECTRICAL

This laboratory has demonstrated its technical competence to operate in accordance with MS
ISO/IEC 17025:2005 (ISO/IEC 17025:2005).
This laboratory’s fulfillment of the requirements of ISO/IEC 17025 means the laboratory
meets both the technical competence requirements and management system requirementsthat are necessary for it to consistently deliver technically valid test results and calibrations.
The management system requirements in ISO/IEC 17025 are written in language relevant tolaboratory operations and operate generally in accordance with the principles of ISO 9001
(see Joint ISO-ILAC-IAF Communiqué dated April 2017).
* The expanded uncertainties are based on an estimated confidence probability ofapproximately 95% and have a coverage factor of k=2 unless stated otherwise.

SCOPE OF CALIBRATION: ELECTRICAL

Instrument Calibrated/
Measurement
Parameter

Range

Calibration andMeasurement Capability
Expressed as
an Uncertainty(±)*-0.1 mV to -100 mV-100.1 mV to -1 V
-1.1 V to -10 V-10.1 V to -100 V-100.1 V to -1000 V

0.0076 mV
0.000025 V
0.00018 V
0.0045 V
0.045 V

0.1 mV to 100 mV
100.1 mV to 1 V
1.1 V to 10 V
10.1 V to 100 V
100.1 V to 1000 V

0.0076 mV
0.000025 V
0.00018 V0.0045 V
0.045 V

0.1 µA to 1 µA
1.1 µA to 10 µA
10.1 µA to 100 µA
100.1 µA to 1 mA
1.1 mA to 10 mA
10.1 mA to 100 mA
100.1 mA to 1 A
1.1 A to 10 A

0.00055 µA
0.0052 µA
0.051 µA
0.00055 mA
0.0070 mA
0.055 mA
0.00090 A
0.014 A

Remarks

SOURCESa) DC Voltage(b) DC Current

SKIM AKREDITASI MAKMAL MALAYSIA (SAMM)
LABORATORY ACCREDITATION SCHEME OF MALAYSIA

Measurement
Using Multimeter
Keysight 34470A

Measurement
Using Multimeter
Keysight 34470A

Schedule
Issue date: 19 July 2017
Valid until: 19 July 2020
MS ISO/IEC 17025

NO: SAMM 839

Page: 2 of 3

SCOPE OF CALIBRATION: ELECTRICAL

Instrument Calibrated/Measurement
Parameter

Scan this QR Code or visit www.jsm.gov.my/cab-directories for the current scope of accreditation(c) AC Voltage

Range
40Hz to 50kHz
10.1 mV to 100 mV
100.1 mV to 1 V
1.1 V to 10 V
10.1 V to 100 V
100.1 V to 750 V(d) AC Current

Calibration andMeasurement Capability
Expressed asan Uncertainty(±)*

0.11 mV
0.0020 V
0.011 V
0.10 V
0.92 V20Hz to 5kHz

Remarks
Measurement
Using Multimeter
Keysight 34470A

Measurement
Using Multimeter
Keysight 34470A

10µA to 100 µA
100.1 µA to 1 mA
1.1 mA to 10 mA
10.1 mA to 100 mA
100.1 mA to 1 A
1.1 A to 10 A

0.15 μA
0.0015 mA
0.0014 mA
0.14 mA
0.0015 A
0.021 A(e) Resistance 2
wire

0.1Ω to 100 Ω
100.1Ω to 1 kΩ
1.1kΩ to 10 kΩ
10.1kΩ to 100 kΩ
100.1kΩ to 1 MΩ
1.1MΩ to10 MΩ
10.1MΩ to 100 MΩ

0.10 Ω
0.00010 kΩ
0.00046 kΩ
0.0045 kΩ
0.000076 MΩ0.0026 MΩ
0.30 MΩ

Measurement
Using Multimeter
Keysight 34470Af) Resistance 4wire0.1Ω to 100 Ω
100.1Ω to 1 kΩ
1.1kΩ to 10 kΩ
10.1kΩ to 100 kΩ
100.1kΩ to 1 MΩ
1.1MΩ to10 MΩ

0.10 Ω
0.00010 kΩ
0.00046 kΩ
0.0045 kΩ
0.000076 MΩ
0.0026 MΩ

Measurement
Using Multimeter
Keysight 34470A

SKIM AKREDITASI MAKMAL MALAYSIA (SAMM)
LABORATORY ACCREDITATION SCHEME OF MALAYSIA

Schedule
Issue date: 19 July 2017
Valid until: 19 July 2020
MS ISO/IEC 17025

NO: SAMM 839

Page: 3 of 3

SCOPE OF CALIBRATION: ELECTRICAL

Instrument Calibrated/Measurement
Parameter

Range

Calibration andMeasurement Capability
Expressed as
an Uncertainty(±)*-0.1 mV to -100 mV-100.1 mV to -1 V
-1.1 V to -10 V-10.1 V to -100 V-100.1 V to -1000 V

0.0013 mV
0.000010 V
0.000040 V
0.00056 V
0.0071 V

0.1 mV to 100 mV
100.1 mV to 1 V
1.1 V to 10 V
10.1 V to 100 V
100.1 V to 1000 V

0.0013 mV
0.000010 V
0.000040 V
0.00056 V
0.0071 V-10µA to -200 µA-200.1 µA to -2mA- 2.1 mA to -20 mA-20.1 mA to -200 mA-200.1 mA to -2 A
-2.1 A to -10 A

0.0015 µA
0.0015 mA
0.00076 mA
0.0099 mA
0.00018 A
0.0041 A

10 µA to 200 µA
200.1 µA to 2 mA
2.1 mA to 20 mA
20.1 mA to 200 mA
200.1 mA to 2 A
2.1 A to 10 A

0.0015 μA
0.0015 mA
0.00076 mA
0.0099 mA
0.00018 A
0.0041 A

Remarks

Scan this QR Code or visit www.jsm.gov.my/cab-directories for the current scope of accreditation

2. INDICATING
INSTRUMENTS(a) DC Voltage(b) DC Current(c) AC Voltage

40Hz to 20kHz
1.1mV to 20 mV
20.1mV to 200 mV
200.1mV to 2 V
2.1 V to 20V
20.1 V to 200 V

0.0060 mV
0.024 mV
0.00012 V
0.0012 V
0.013 V

40Hz to 1kHz
200.1 V to 1000 V

0.085 VSKIM AKREDITASI MAKMAL MALAYSIA (SAMM)
LABORATORY ACCREDITATION SCHEME OF MALAYSIA

Generating using
Calibrator model
Fluke 5700A

Generating using
Calibrator model
Fluke 5700A &Fluke 5725A
Amplifier

Generating using
Calibrator model
Fluke 5700A

Schedule
Issue date: 19 July 2017
Valid until: 19 July 2020
MS ISO/IEC 17025

NO: SAMM 839

Page: 4 of 3

SCOPE OF CALIBRATION: ELECTRICAL

Instrument Calibrated/Measurement
Parameter

Scan this QR Code or visit www.jsm.gov.my/cab-directories for the current scope of accreditation(d) AC Currente) Resistance

Range
40Hz to 10kHz
10 µA to 200 µA
200.1 µA to 2 mA
2.1 mA to 20 mA
20.1 mA to 200 mA
200.1 mA to 2 A
2.1 A to 10 A

Calibration andMeasurement Capability
Expressed as
an Uncertainty(±)*
0.30 μA
0.0029 mA
0.028 mA
0.24 mA
0.015 A
0.037 AFIXED VALUE
1Ω
10 Ω
100 Ω
1 kΩ
10 kΩ
100 kΩ
1 MΩ
10 MΩ
100 MΩ

0.000098 Ω0.000091 Ω
0.00026 Ω0.0000088 kΩ
0.000020 kΩ
0.00023 kΩ
0.000021 MΩ
0.00014 MΩ
0.0045 MΩ

SKIM AKREDITASI MAKMAL MALAYSIA (SAMM)
LABORATORY ACCREDITATION SCHEME OF MALAYSIA

Remarks
Generating using
Calibrator model
Fluke 5700A &Fluke 5725A
Amplifier

Generating using
Calibrator model
Fluke 5700A

ScheduleIssue date: 19 July 2017
Valid until: 19 July 2020
MS ISO/IEC 17025

NO: SAMM 839

Page: 5 of 3

SCOPE OF CALIBRATION: ELECTRICAL
SITE: CATEGORY I
Instrument Calibrated/Measurement
Parameter

Range

Calibration andMeasurement Capability
Expressed as
an Uncertainty(±)*-0.1 mV to -100 mV-100.1 mV to -1 V
-1.1 V to -10 V-10.1 V to -100 V-100.1 V to -1000 V

0.0076 mV
0.000025 V
0.00018 V
0.0045 V
0.045 V

0.1 mV to 100 mV
100.1 mV to 1 V
1.1 V to 10 V
10.1 V to 100 V
100.1 V to 1000 V

0.0076 mV
0.000025 V
0.00018 V
0.0045 V
0.045 V100.1 nA to 1 µA
1.1 µA to 10 µA
10.1 µA to 100 µA
100.1 µA to 1 mA
1.1 mA to 10 mA
10.1 mA to 100 mA
100.1 mA to 1 A
1.1 A to 10 A
40Hz to 50kHz

0.00055 µA
0.0052 µA
0.051 µA
0.00055 mA
0.0070 mA
0.055 mA
0.00090 A
0.014 A

10.1 mV to 100 mV
100.1 mV to 1 V
1.1 V to 10 V
10.1 V to 100 V
100.1 V to 750 V

0.11 mV
0.0020 V
0.011 V
0.10 V
0.92 V

Remarks

Scan this QR Code or visit www.jsm.gov.my/cab-directories for the current scope of accreditation

1. SOURCES(a) DC Voltage(b) DC Current(c) AC Voltage(d) AC Current

20Hz to 5kHz
10µA to 100 µA
100.1 µA to 1 mA
1.1 mA to 10 mA
10.1 mA to 100 mA
100.1 mA to 1 A
1.1 A to 10 A

0.15 μA
0.0015 mA
0.0014 mA
0.14 mA
0.0015 A
0.021 ASKIM AKREDITASI MAKMAL MALAYSIA (SAMM)
LABORATORY ACCREDITATION SCHEME OF MALAYSIA

Measurement
Using Multimeter
Keysight 34470A

Measurement
Using Multimeter
Keysight 34470A

Measurement
Using Multimeter
Keysight 34470AMeasurement
Using Multimeter
Keysight 34470A

Schedule
Issue date: 19 July 2017
Valid until: 19 July 2020
MS ISO/IEC 17025

NO: SAMM 839

Page: 6 of 3

SCOPE OF CALIBRATION: ELECTRICAL
SITE: CATEGORY I
Instrument Calibrated/
Measurement
Parameter

Scan this QR Code or visit www.jsm.gov.my/cab-directories for the current scope of accreditation(e) Resistance 2
wire(f) Resistance 4
wire

0.1Ω to 100 Ω
100.1Ω to 1 kΩ
1.1kΩ to 10 kΩ
10.1kΩ to 100 kΩ
100.1kΩ to 1 MΩ
1.1MΩ to10 MΩ
10.1MΩ to 100 MΩ
0.1Ω to 100 Ω
100.1Ω to 1 kΩ
1.1kΩ to 10 kΩ
10.1kΩ to 100 kΩ
100.1kΩ to 1 MΩ
1.1MΩ to10 MΩ

Calibration andMeasurement Capability
Expressed as
an Uncertainty(±)*
0.10 Ω
0.00010 kΩ
0.00046 kΩ
0.0045 kΩ
0.000076 MΩ
0.0026 MΩ
0.30 MΩ
0.10 Ω
0.00010 kΩ
0.00046 kΩ
0.0045 kΩ
0.000076 MΩ
0.0026 MΩ

-0.1 mV to -100 mV-100.1 mV to -1 V-1.1 V to -10 V-10.1 V to -100 V-100.1 V to -1000 V

0.0013 mV
0.000010 V
0.000040 V
0.00056 V
0.0071 V

0.1 mV to 100 mV
100.1 mV to 1 V
1.1 V to 10 V
10.1 V to 100 V
100.1 V to 1000 V-10µA to -200 µA-200.1 µA to -2mA
- 2.1 mA to -20 mA-20.1 mA to -200 mA-200.1 mA to -2 A
-2.1 A to -10 A

0.0013 mV
0.000010 V
0.000040 V
0.00056 V
0.0071 V
0.0015 µA
0.0015 mA
0.00076 mA
0.0099 mA
0.00018 A
0.0041 A

10 µA to 200 µA
200.1 µA to 2 mA
2.1 mA to 20 mA
20.1 mA to 200 mA
200.1 mA to 2 A
2.1 A to 10 A

0.0015 μA
0.0015 mA
0.00076 mA
0.0099 mA
0.00018 A
0.0041 A

Range

Remarks
Measurement
Using Multimeter
Keysight 34470A

Measurement
Using Multimeter
Keysight 34470A

2. INDICATING
INSTRUMENTS(a) DC Voltage(b) DC Current

SKIM AKREDITASI MAKMAL MALAYSIA (SAMM)
LABORATORY ACCREDITATION SCHEME OF MALAYSIA

Generating using
Calibrator model
Fluke 5700A

Generating using
Calibrator model
Fluke 5700A &Fluke 5725A
AmplifierSchedule
Issue date: 19 July 2017
Valid until: 19 July 2020
MS ISO/IEC 17025

NO: SAMM 839

Page: 7 of 3SCOPE OF CALIBRATION: ELECTRICAL
SITE: CATEGORY I
Instrument Calibrated/Measurement
Parameter

Scan this QR Code or visit www.jsm.gov.my/cab-directories for the current scope of accreditation(c) AC Voltage( d ) AC Current

Range

Calibration andMeasurement Capability
Expressed as
an Uncertainty(±)*

40Hz to 20kHz
1.1mV to 20 mV
20.1mV to 200 mV
200.1mV to 2 V
2.1 V to 20V
20.1 V to 200 V

0.0060 mV
0.024 mV
0.00012 V
0.0012 V
0.013 V

40Hz to 1kHz
200.1 V to 1000 V

0.085 V

40Hz to 10kHz
10 µA to 200 µA
200.1 µA to 2 mA
2.1 mA to 20 mA
20.1 mA to 200 mA
200.1 mA to 2 A
2.1 A to 10 A

0.30 μA
0.0029 mA
0.028 mA
0.24 mA
0.015 A
0.037 AFIXED VALUEe) Resistance

1Ω
10 Ω
100 Ω
1 kΩ
10kΩ
100 kΩ
1 MΩ
10 MΩ
100 MΩ

0.000098 Ω
0.000091 Ω
0.00026 Ω
0.0000088 kΩ
0.000020 kΩ
0.00023 kΩ
0.000021 MΩ
0.00014 MΩ
0.0045 MΩ

Signatories:
1.
2.

Ng Kah Wai
Shazrul Azmir

SKIM AKREDITASI MAKMAL MALAYSIA (SAMM)
LABORATORY ACCREDITATION SCHEME OF MALAYSIA

Remarks
Generating using
Calibrator model
Fluke 5700A

Generating using
Calibrator model
Fluke 5700A &Fluke 5725A
Amplifier

Generating using
Calibrator model
Fluke 5700A