DS252PAPERLESS RECORDER‫ە‬Code-table function
The code can use alphanumeric characters and/or symbols for up to6 characters or digits. Up to 100 codes can be classified and tallied
out in each channel.

‫ە‬High-level statistical calculation function
DS252 is pre-loaded with a function that calculates grand total,sub-total, maximum, minimum, average or standard deviation of therecorded data.At the same time, DS252 is able to grasp the extent ofdata variation using its frequency distribution function.
DIN96‫ە‬Simple operation & code-table registration via PCDS252 codes can be saved in the CF Card. And reversely, codes savedin the CF card can be written in DS252 internal memory.

‫ە‬Optional interfaces‫ە‬Mass recording in CF card
Weight /load data of indicator or DS252 statistical data can be
mass-recorded in CF Card.The data equivalent to amount of data
printable on 400 rolls ͤof rolled paper can be recorded in 128MB CFcard.Even without a CF Card, The data equivalent to amount of data
printable on 1 roll ͤor more rolled paper can be recorded in DS252
internal memory.
ͤIn comparison with Unipulse printer M252B(which is able to print 8000 rows on 1 roll of rolled paper)

Basically, SI/F interfaces are installed in 2 channels to enable theeconomical double channels recordings. However, as option interfaces,
there are RS-232C and BCD input interfaces available.
With that, data from various kinds of devices can be recorded.

‫ڦ‬Application Example :Recording code data of an option and weight data of an indicator

RS-232C or
BCD input

SI/F AchIndicator (Weight data)
SI/F Bch

PLC (Ach Code data)
Record data is
CSV format.

Structure of product code

Indicator (Weight data)By entering via the front panel key, weight data with code can berecorded in both A ch and B ch. Code data of A ch can also be entered
via an optional interface.

Specifications①Standard unit

②Interface
SignStandard

CARD SLOTInterface
SI/F (2ch)↓ One optional interface can be added in addition the standard interface.

232
BCI

RS-232C
BCD input

Card specification

CF-Card (Type I or Type II)Power supply

5V

Available capacity

Card Max. 512 MB (FAT16)

File format

CSV Format Text File

File nameddhhmmss.csv (dd: day, hh: hour, mm: minute, ss: second)

File writing method

WRITEЍENT key input
GTЍENT key input
Rear panel input (WRITE terminal)
Rear panel input (GT terminal)

External dimension(Front View)

In case of Full Internal Memory
Write when the pre-set recording frequency is exceeded(Side View)
DISPLAY

Setting time
Black & white LCD display of 16 digits 4.3mm height characters at 2 rows

FUNCTIONS

Data recording

REC key inputmethod

Rear panel input (REC terminal)
Auto Recording command input (via SI/F, RS-232C)

96

Automatic recording based interval setting (1 – 9999 sec)
Details of recorded

Date, Time, Input value, Unit, Status, Code name (6-digits),

dataCount (1 to 9999 counts), Sub-total (9-digits), Grand Total (9-digits),
Batch Total, Maximum Value,Minimum Value, Average,Standard
Deviation, Range (Max - Min), Frequency Distribution, Error, OverINTERFACE

SIF: 2-wire type serial interface (2 ch provided as standard)
BCI: BCD Parallel data input interface (Option)
232: RS-232C communication interface (Option)

1096

100

Only one option can be installed.
GENERAL
Power supply voltage DC24V±15㸣
SPECIFICATIONS Power consumptionMax. 150 mA(Rear View)

Inrush current

10A (800ȣsec(Typ))Operation condition

Temperature: Operation Temperature Range: 0Υ to +40ΥExternal dimension

Humidity: 85% RH or less (non-condensing)

Weight96(W)×96(H)×110(D)mm (not including protrusions)91

Panel cut dimension
Panel thickness 1.6 to 3.2mm

92 +1-0

Storage Temperature Range: -20Υ to +70Υ

ATTACHMENTS

App. 0.8 kg
Mini driver ͐͐͐͐͐͐͐͐͐͐͐͐͐͐͐͐͐͐͐͐͐͐͐͐͐
BCD input connector (When BCD input option is selected) ͐͐͐͐OPTIONALACCESSORIES

91

92 +1-0Unit mm

Operation Manual ͐͐͐͐͐͐͐͐͐͐͐͐͐͐͐͐͐͐͐͐͐͐
CF128M:
CF Card 128 MByte
CF512M:

CF Card 512 MByte

CF-ADP:

CF Card adapter (ATA TYPE II)