DSCBHIGH STIFFNESS DIGITAL LOAD CELL

Detachable cable

Flexible cable

Measurement is possible by easily connecting to PC!
Hard-to-break force measurement tool capable ofstable measurement of even severe load!
High stiffness & safe overload of 500%

Excellent maintainability

As deformation under load is small, DCSB can be used safely
for cutting by machine tools or for measuring grinding force.
With its 500% safe overload, DSCB can stably measure evensevere load over a long period.

Zero point and full scale are adjusted at the time of shipmentfrom a factory; recalibration is unnecessary at a replacement ofDSCB. As a cable and the main unit are connected with aconnector, troublesome cable laying operations at areplacement of the main unit is unnecessary. Misalignment ofzero point from a time of shipment to the present can bechecked, thus abnormality can be promptly detected.

Noise resistant digital output

Abundant functions incorporated

Output format is compliant with RS-485 standard thus is noiseresistant; can be installed with long-distance cable.

HI/LO limit comparison & a wide variety of hold functions areincluded.
Digital low-pass filter and moving average filter can be adjusted
by commands.

Connect to PC and PLC directly without dedicated indicator
By connecting to PC, draw waveforms, logging and such operations can be achieved with dedicated software (for free).
● OK/NOK & stability judgement function
Upper/lower limit comparators (HH/ HI/ OK/ LOvLL)● Easy initial setting
Export/Import of setting values

USB dedicated software for DSCB● Record
Record measured values and status (such as upper/lower limit comparison)

Display load variation on a load cell in real time via USB.

Display setting

Display load

● Waveform drawing
By real-time waveform display, visualizes load change of a load cell
Stability & upper/lower limit comparison status can be checked in one glance
Waveform data can be recorded■ Display setting
Waveform can be acquired up to acquisition time after<Normal>Start
button is clicked.
Data acquisition resumes when Stop button is clicked.
〇Continuous data recording function
Continue data recording until〇USB dedicated software for DSCB<History>

Stopbutton is clicked.

Once Start button is clicked, the status will turn to trigger standby.
Data acquisition starts when data level passes the specified value
in the specified direction.
Display past data in waveform■ Trigger function- Pre-trigger- Level trigger- Trigger slope* RS-485-USB converter is required to use the USB dedicated software for DSCB.
Recommended product: USB-4W485i-RJ45-T9P (Made by System Sacom Industry Corp.)

Example of use

Specifications
DSCB-1KN DSCB-2KN DSCB-5KN DSCB-10KN DSCB-20KN DSCB-50KN

Model1 kN

Rated capacity

2 kN

5 kN

10 kN

Safe overload

500% R.C.

Non-linearity

0.1% or less R.O.

Hysteresis
Repeatability

0.2% or less R.O.

20 kN

Press machine

50 kN

0.1% or less R.O.

Compensated temperature range

PLC-10 to +60

Safe temperature range-10 to +70

Temperature effect on zero

0.015% R.O/

or belowTemperature effect on span

0.01% R.O/

or below

Converter resolution

DSCB

RS-485

RS-485 from/to USB converter

TORQUE METER

Loadcell material

Stainless steel

Power supply voltage

DC 12 to 24 V±15% (Insulated internally)

Power consumption

0.7 W or less

Digital low-pass filter

Can be set anywhere between OFF, 0.1 to 999.9 Hz

Moving average filter
Sampling rate

Can be set anywhere between OFF, 2 to 999 times

Interface
Operation & setting methods

Display digit number

DC 12 to 24 V

Cable

4800 times/sec.
RS-485 interface (4-wire)
Display & configuration by command operation or through dedicatedsoftware (Free software)

5 digits (-99999 to +99999 plus decimal point)- Digital zero
- OK/NOK judgement (Upper and lower limit comparison: HH/ HI/ OK/ LO/ LL)- Hold function (Sample. Peak, Bottom, Peak & Bottom, Average)- Calibration date / rated capacity / serial info read-out

Other functions

Flexible cableConnector (plug)
LF07WBP-6S(31)- Indicated value display, - Waveform display- Setting changes, - Data logging function

Dedicated software function

PC

Recommended product:USB-4W485i-RJ45-T9P(Made by System Sacom Industry Corp.)

24 bitLOAD CELL

Cable

5 6-conductor flexible cable 2 m, Cable end: 7 wires

Accessories

5 6-conductor flexible cable 5 m, Cable end: 7 wires

App. 10 cm2 m*

1: RedPWR (+12 to +24 V)
2: Yellow RS-485 TX+
3: Orange RS-485 TX4: Black PWR (0 V)
5: Green RS-485 RX+
6: White RS-485 RXSHIELD* Cable length can be extended to 5 m(Optional)

EMC directives EN61326-1, EN61326-2-3

CE marking certification

External dimension

DSCB-1KN/2KN/5KN

DSCB-10KN/20KN
Load direction(+)

SR50(3)

1012

SR75

2522

CLAMPING FORCEMEASUREMENT

2522(3)

Load direction(+)

Approx. 36

60

Approx. 36

60

9 0°

9 0°

P.C.D.52.6

P.C.D.48°°

22.5

22.54-M4 Depth 6.5

5 6-conductor flexible cable 2 m
Cable end: 7 wires*

5 6-conductor flexible cable 2 m
Cable end: 7 wires*
4-M4 Depth 6

FORCE
MEASUREMENTDSCB-50KN
Load direction(+)(5)

24SR15040

35* Connector can be attached to the cable according to your requests.
Refer to 82 pages for details.

Approx. 36
100

P.C.D.80

Unit: mm

2 2.5°

WEIGHING
MEASUREMENT

9 0°

5 6-conductor flexible cable 2 m
Cable end: 7 wires*
4-M

42 DSCB5D

ep

th 8Model
DSCB-1KN
DSCB-2KN
DSCB-5KN
DSCB-10KN
DSCB-20KN
DSCB-50KN

Deflection at rated0.010 mm
0.012 mm0.021 mm0.025 mm0.035 mm0.065 mmNatural frequency
22 kHz
22 kHz21 kHz24 kHz
26 kHz
15 kHzWeight
300 g
300 g
400 g
400 g
400 g
2.0 kg