UHPLC & HPLC HILIC column
SPE

新世代ヒリックカラム

Website: www.hilicon.com

株式会社 クロマニックテクノロジーズ

4種類の固定相を提供

iHILIC®-(P) Classic
Polymer based

iHILIC-Fusion(P) and iHILIC-(P) Classic: pH 1-10; 5 μm

iHILIC®-(P) Classic is capable of the equivalent separation of ZIC-pHILIC.
ヌクレオチドの分離

人工甘味料の分離

iHILIC®-Fusion
AMP

iHILIC®-Fusion(+)

ADP
ATP

NaphthaleneiHILIC®-Fusion(+)
AMP
ADP

Naphthalene

ATP

Column: 100 × 2.1 mm, 3.5-µm 100ÅMobile phase: A) 100 mM ammonium formate,
pH = 5.8 / acetonitrile = 30 /70
Flow rate: 0.2 mL/min
Column temperature: Ambient
Injection volume: 2 µL

LC–MS–MS system: Advance UHPLC with a CTC
PAL autosampler and Bruker EVOQ Elite triple
quadrupole system with electrospray ionization;
multiple reaction monitoring mode (MRM) in ESI(+)
and ESI(-) was used for selective and sensitive
detection, see details in Table.
Column: 150 × 2.1 mm, 3.5-µm 100ÅiHILIC-Fusion(+)
Gradient elution: A) acetonitrile; B) 10 mM
ammonium formate, pH = 3.5; gradient elution
from (95/5) A/B to (84/16) A/B in 8.5 min.
Flow rate: 0.3 mL/min
Column temperature: 40 °C
Injection volume: 5 µL

アミノ酸の分離標準品
iHILIC®-Fusion(+)

ダイエットサプリ

LC–MS System: Agilent 1100er LC system and
Thermo Fisher LTQ equipped with a HESI source,
operated in positive ionization mode for analysis ofstandards. For the dietary supplement, an
Orbitrap Exactive classic equipped with a HESI
source and operated in positive ionization mode.
Column: 150 × 2.1 mm, 3.5-μm iHILIC-Fusion(+)
Gradient Elution: A) acetonitrile–water–1 M
ammonium acetate, pH 5.75 (90:5:5); B) water–acetonitrile–1 M ammonium acetate, pH 5.75
(90:5:5); 0–0.5 min (90:10) A–B; 0.5 to 25 min,gradient elution from (90:10) A–B to (60:40) A–B.
Flow Rate: 0.3 mL/min
Column Temperature: 40 °C
Injection Volume: 5 μL
Amino Acids: Arginine, asparagine, aspartic acid,glutamic acid, glutamine, hydroxyl-proline,
isoleucine, leucine, lysine, phenylalanine, proline,
tryptophan, tyrosine, and valine. 50 μM of eachamino acid was dissolved in water–acetonitrile(25:75) solution.

リン脂質の分離リゾビスフォスファチジン酸

ホスファチジン酸

ホスファチジルグリセロール

iHILIC®-Fusion

カルジオリピンホスファチジルイノシトール

ホスファチジルコリン

ホスファチジルエタノールアミン

リゾホスファチジルコリン

ホスファチジルセリン

リゾホスファチジン酸

リン脂質の化学構造

LC–MS system: Thermo Scientifi c Ultimate 3000 HPLC
system coupled to a Q Exactive Plus Hybrid
Quadrupole-Orbitrap mass spectrometer, which isequipped with a HESI II source and operated in negative
ionization mode (ESI-).
Column: 150 × 2.1 mm, 3.5 μm 100 Å iHILIC -Fusion.
Eluent:A) ammonium acetate solution (35 mM, pH 5.75) and
acetonitrile (95:5, v/v);
B) acetonitrile.
Gradient elution: 0–0.5 min, 97% B; 0.5–26.5 min, from
97% to 75% B; 27–33 min, 60% B; 35–45 min 97% B.
Flow rate: 0.3 mL/min
Column temperature: 40 °C
Injection volume: 10 μL®

サイロシンとサイロシビンの分離

Experimental
LC–MS System: Agilent 1100 LC system andBruker Esquire 6000 ion trap mass spectrometer,
operated in positive ionization mode (ESI+).
Chromatographic data were acquired andevaluated with ChemStation Rev. A. 10.02.
Column: 150 × 4.6 mm, 3.5-μm 100 Å iHILIC®Fusion (P/N 110.154.0310, HILICON AB, Sweden)
Mobile Phase: 80:20 (v/v) acetonitrile–ammonium
format (10 mM, pH 3.5)
Flow Rate: 0.5 mL/min
Column Temperature: 12 °C
Sample Preparation: Quasi-counter current
extraction with methanol at 60 °C in a Shimadzu
10/A HPLC system. A 50- mg measure of air-dried
and homogenized hallucinogenic mushroom wasfilled in the extractor chamber (an empty 250 ×
4.6 mm HPLC column). The standard solutions
were 5 μg/ mL and 500 μg/mL for psilocin andpsilocybin, respectively. Methanol was used as thesolvent.
Injection Volume: 1 μL

有機リン酸エステルの分離

Experimental
LC–MS system 1: Thermo Ultimate 3000 LC system and AB SCIEX 3200 QTrapR equipped with an ESI source, operated
in ESI(+) and ESI(-) mode for analysis of standards. Column: 150 × 2.1 mm, 3.5-μm iHILIC -Fusion(+) (P/N 100.152.0310,
HILICON AB). Gradient elution 1: A) acetonitrile; B) deionized water-1 M ammonium formate, pH 2.9 (99.5:0.5); 0–3.5
min (90:10) A–B; 3.5 to 6.5 min, gradient elution from (90:10) A–B to (50:50) A–B; 6.5–14.5 min, (50:50) A–B.
Flow rate: 0.3 mL/min. Column temperature: 40 °C. Injection volume: 10 μL.
®

輸入元

販売元

株式会社バイオニック

株式会社 クロマニックテクノロジーズ

417-0801 富士市大渕3397-19
TEL: 0545-38-9125 FAX: 0545-38-9126
Email: info@bionikinc.com URL: http:// www.bionikinc.com

552-0001 大阪府大阪市港区波除6-3-1
TEL: 06-6581-0885 FAX: 06-6581-0890
E-mail: info@chromanik.co.jp URL: http://chromanik.co.jp2104

UHPLC & HPLC HILIC column
SPE

価格表
2024年2月より

iHILIC® HILIC Columns in UHPLC and HPLC
iSPE®-HILIC - SPE Cartridges and 96-Well Plates

HILICON AB社製カラム
Website: www.hilicon.com

クロマニックテクノロジーズ

iHILIC-Fusion(P) and iHILIC-(P) Classic: pH 1-10; 5 μm

iHILIC®-Fusion - UHPLC HILIC Column in SS
P/N
110.022.0110
110.032.0110
110.052.0110
110.102.0110
110.152.0110
110.053.0110
110.103.0110
110.153.0110

Product NameiHILIC®-Fusion, HILIC Column, SS, 20x2.1mm, 1.8µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 30x2.1mm, 1.8µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 50x2.1mm, 1.8µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 100x2.1mm, 1.8µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 150x2.1mm, 1.8µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 50x3.0mm, 1.8µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 100x3.0mm, 1.8µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 150x3.0mm, 1.8µm, 100Å

価格
¥144,000¥147,000¥163,000¥175,000¥203,000¥166,000¥193,000¥221,000

111.122.0110
111.222.0110

iHILIC®-Fusion, Guard Column, SS, 20x2.1mm, 1.8µm, 100Å, 1pk
iHILIC®-Fusion, Guard Column, SS, 20x2.1mm, 1.8µm, 100Å, 2pk*

¥92,000¥203,000iHILIC®-Fusion - UHPLC HILIC Column in PEEK-SS
P/N
140.051.0110
140.101.0110
140.151.0110
140.052.0110
140.102.0110
140.152.0110

Product NameiHILIC®-Fusion, HILIC Column, PEEK-SS, 50x1mm, 1.8µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK-SS, 100x1mm, 1.8µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK-SS, 150x1mm, 1.8µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK-SS, 50x2.1mm, 1.8µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK-SS, 100x2.1mm, 1.8µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK-SS, 150x2.1mm, 1.8µm, 100ÅiHILIC®-Fusion - HPLC HILIC Column in SS
価格¥230,000¥252,000¥273,000¥255,000¥267,000¥285,000

SS - Stainless steel column hardware
PEEK-SS - PEEK lined stainless steel column hardware* One UHPLC column coupler is included

iHILIC®-Fusion - HPLC HILIC Column in PEEK
P/N
110.022.0310
110.032.0310
110.052.0310
110.102.0310
110.152.0310
110.054.0310
110.104.0310
110.154.0310
110.254.0310

Product NameiHILIC®-Fusion, HILIC Column, PEEK, 20x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK, 30x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK, 50x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK, 100x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK, 150x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK, 50x4.6mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK, 100x4.6mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK, 150x4.6mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK, 250x4.6mm, 3.5µm, 100Å

価格
¥132,000¥135,000¥144,000¥154,000¥190,000¥157,000¥175,000¥212,000¥261,000

110.022.0510
110.032.0510
110.052.0510
110.102.0510
110.152.0510
110.252.0510
110.054.0510
110.104.0510
110.154.0510
110.254.0510

iHILIC®-Fusion, HILIC Column, PEEK, 20x2.1mm, 5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK, 30x2.1mm, 5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK, 50x2.1mm, 5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK, 100x2.1mm, 5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK, 150x2.1mm, 5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK, 250x2.1mm, 5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK, 50x4.6mm, 5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK, 100x4.6mm, 5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK, 150x4.6mm, 5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK, 250x4.6mm, 5µm, 100Å

¥129,000¥135,000¥141,000¥151,000¥187,000¥221,000¥154,000¥169,000¥203,000¥245,000

111.122.0510
111.322.0510

iHILIC®-Fusion, Guard Column, PEEK, 20x2.1mm, 5µm, 100Å, 1pk
iHILIC®-Fusion, Guard Column, PEEK, 20x2.1mm, 5µm, 100Å, 3pk*

¥83,000¥203,000

PEEK - Poly(ether ether ketone) engineering plastic column hardware
SS - Stainless steel column hardware
PEEK-SS - PEEK-lined stainless steel column hardware* One HPLC column coupler is included

P/N
114.022.0310
114.032.0310
114.052.0310
114.102.0310
114.152.0310
114.252.0310
114.053.0310
114.103.0310
114.153.0310
114.253.0310
114.054.0310
114.104.0310
114.154.0310
114.254.0310

Product NameiHILIC®-Fusion, HILIC Column, SS, 20x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 30x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 50x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 100x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 150x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 250x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 50x3mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 100x3mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 150x3mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 250x3mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 50x4.6mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 100x4.6mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 150x4.6mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 250x4.6mm, 3.5µm, 100Å

価格
¥144,000¥144,000¥151,000¥166,000¥184,000¥233,000¥154,000¥166,000¥196,000¥239,000¥163,000¥181,000¥227,000¥291,000

114.022.0510
114.032.0510
114.052.0510
114.102.0510
114.152.0510
114.252.0510
114.053.0510
114.103.0510
114.153.0510
114.253.0510
114.054.0510
114.104.0510
114.154.0510
114.254.0510

iHILIC®-Fusion, HILIC Column, SS, 20x2.1mm, 5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 30x2.1mm, 5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 50x2.1mm, 5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 100x2.1mm, 5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 150x2.1mm, 5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 250x2.1mm, 5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 50x3mm, 5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 100x3mm, 5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 150x3mm, 5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 250x3mm, 5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 50x4.6mm, 5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 100x4.6mm, 5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 150x4.6mm, 5µm, 100ÅiHILIC®-Fusion, HILIC Column, SS, 250x4.6mm, 5µm, 100Å

¥144,000¥144,000¥147,000¥163,000¥178,000¥227,000¥147,000¥160,000¥184,000¥227,000¥157,000¥172,000¥215,000¥273,000iHILIC®-Fusion - HPLC HILIC Column in PEEK-SS
P/N
140.051.0310
140.101.0310
140.151.0310
140.052.0310
140.102.0310
140.152.0310
140.252.0310
140.054.0310
140.104.0310
140.154.0310
140.254.0310

Product NameiHILIC®-Fusion, HILIC Column, PEEK-SS, 50x1.0mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK-SS, 100x1.0mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK-SS, 150x1.0mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK-SS, 50x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK-SS, 100x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK-SS, 150x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK-SS, 250x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK-SS, 50x4.6mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK-SS, 100x4.6mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK-SS, 150x4.6mm, 3.5µm, 100ÅiHILIC®-Fusion, HILIC Column, PEEK-SS, 250x4.6mm, 3.5µm, 100Å

価格
¥224,000¥236,000¥252,000¥245,000¥255,000¥267,000¥291,000¥252,000¥270,000¥288,000¥325,000※上記価格は2024年2月1日から適用されます。
※上記価格には消費税は含まれておりません。

iHILIC®-Fusion(+) - UHPLC HILIC Column in SS

iHILIC®-Fusion(+) - HPLC HILIC Column in SS

P/N
100.022.0110
100.032.0110
100.052.0110
100.102.0110
100.152.0110
100.053.0110
100.103.0110
100.153.0110

Product NameiHILIC®-Fusion(+), HILIC Column, SS, 20x2.1mm, 1.8µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 30x2.1mm, 1.8µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 50x2.1mm, 1.8µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 100x2.1mm, 1.8µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 150x2.1mm, 1.8µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 50x3.0mm, 1.8µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 100x3.0mm, 1.8µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 150x3.0mm, 1.8µm, 100Å

価格
¥144,000¥147,000¥163,000¥175,000¥203,000¥166,000¥193,000¥221,000

101.122.0110
101.222.0110

iHILIC®-Fusion(+), Guard Column, SS, 20x2.1mm, 1.8µm, 100Å, 1pk
iHILIC®-Fusion(+), Guard Column, SS, 20x2.1mm, 1.8µm, 100Å, 2pk*

¥92,000¥203,000iHILIC®-Fusion(+) - UHPLC HILIC Column in PEEK-SS
P/N
150.051.0110
150.101.0110
150.151.0110
150.052.0110
150.102.0110
150.152.0110

Product NameiHILIC®-Fusion(+), HILIC Column, PEEK-SS, 50x1mm, 1.8µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK-SS, 100x1mm, 1.8µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK-SS, 150x1mm, 1.8µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK-SS, 50x2.1mm, 1.8µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK-SS, 100x2.1mm, 1.8µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK-SS, 150x2.1mm, 1.8µm, 100Å

価格
¥230,000¥252,000¥273,000¥255,000¥267,000¥285,000

SS - Stainless steel column hardware
PEEK-SS - PEEK lined stainless steel column hardware* One UHPLC column coupler is included

iHILIC®-Fusion(+) - HPLC HILIC Column in PEEK
P/N
100.022.0310
100.032.0310
100.052.0310
100.102.0310
100.152.0310
100.054.0310
100.104.0310
100.154.0310
100.254.0310

Product NameiHILIC®-Fusion(+), HILIC Column, PEEK, 20x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK, 30x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK, 50x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK, 100x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK, 150x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK, 50x4.6mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK, 100x4.6mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK, 150x4.6mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK, 250x4.6mm, 3.5µm, 100Å

価格
¥132,000¥135,000¥144,000¥154,000¥190,000¥157,000¥175,000¥212,000¥261,000

100.022.0510
100.032.0510
100.052.0510
100.102.0510
100.152.0510
100.252.0510
100.054.0510
100.104.0510
100.154.0510
100.254.0510

iHILIC®-Fusion(+), HILIC Column, PEEK, 20x2.1mm, 5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK, 30x2.1mm, 5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK, 50x2.1mm, 5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK, 100x2.1mm, 5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK, 150x2.1mm, 5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK, 250x2.1mm, 5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK, 50x4.6mm, 5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK, 100x4.6mm, 5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK, 150x4.6mm, 5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK, 250x4.6mm, 5µm, 100Å

¥129,000¥135,000¥141,000¥151,000¥187,000¥221,000¥154,000¥169,000¥203,000¥245,000

101.122.0510
101.322.0510

iHILIC®-Fusion(+), Guard Column, PEEK, 20x2.1mm, 5µm, 100Å, 1pk
iHILIC®-Fusion(+), Guard Column, PEEK, 20x2.1mm, 5µm, 100Å, 3pk*

¥83,000¥203,000

P/N
104.022.0310
104.032.0310
104.052.0310
104.102.0310
104.152.0310
104.252.0310
104.053.0310
104.103.0310
104.153.0310
104.253.0310
104.054.0310
104.104.0310
104.154.0310
104.254.0310

Product NameiHILIC®-Fusion(+), HILIC Column, SS, 20x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 30x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 50x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 100x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 150x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 250x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 50x3mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 100x3mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 150x3mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 250x3mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 50x4.6mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 100x4.6mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 150x4.6mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 250x4.6mm, 3.5µm, 100Å

価格
¥144,000¥144,000¥151,000¥166,000¥184,000¥233,000¥154,000¥166,000¥196,000¥239,000¥163,000¥181,000¥227,000¥291,000

104.022.0510
104.032.0510
104.052.0510
104.102.0510
104.152.0510
104.252.0510
104.053.0510
104.103.0510
104.153.0510
104.253.0510
104.054.0510
104.104.0510
104.154.0510
104.254.0510

iHILIC®-Fusion(+), HILIC Column, SS, 20x2.1mm, 5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 30x2.1mm, 5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 50x2.1mm, 5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 100x2.1mm, 5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 150x2.1mm, 5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 250x2.1mm, 5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 50x3mm, 5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 100x3mm, 5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 150x3mm, 5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 250x3mm, 5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 50x4.6mm, 5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 100x4.6mm, 5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 150x4.6mm, 5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, SS, 250x4.6mm, 5µm, 100Å

¥144,000¥144,000¥147,000¥163,000¥178,000¥227,000¥147,000¥160,000¥184,000¥227,000¥157,000¥172,000¥215,000¥273,000iHILIC®-Fusion(+) - HPLC HILIC Column in PEEK-SS
P/N
150.051.0310
150.101.0310
150.151.0310
150.052.0310
150.102.0310
150.152.0310
150.252.0310
150.054.0310
150.104.0310
150.154.0310
150.254.0310

Product NameiHILIC®-Fusion(+), HILIC Column, PEEK-SS, 50x1.0mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK-SS, 100x1.0mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK-SS, 150x1.0mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK-SS, 50x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK-SS, 100x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK-SS, 150x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK-SS, 250x2.1mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK-SS, 50x4.6mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK-SS, 100x4.6mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK-SS, 150x4.6mm, 3.5µm, 100ÅiHILIC®-Fusion(+), HILIC Column, PEEK-SS, 250x4.6mm, 3.5µm, 100Å

価格
¥224,000¥236,000¥252,000¥245,000¥255,000¥267,000¥291,000¥252,000¥270,000¥288,000¥325,000

PEEK - Poly(ether ether ketone) engineering plastic column hardware
SS - Stainless steel column hardware
PEEK-SS - PEEK-lined stainless steel column hardware* One HPLC column coupler is included※上記価格は2024年2月1日から適用されます。
※上記価格には消費税は含まれておりません。

iHILIC®-(P) Classic
Polymer based
iHILIC®-(P) Classic is capable of the equivalent separation of ZIC-pHILIC.
iHILIC®-Fusion(P) - HPLC HILIC Column in PEEKiHILIC®-(P) Classic - HPLC HILIC Column in PEEK

P/N
130.022.0520
130.032.0520
130.052.0520
130.102.0520
130.152.0520
130.252.0520
130.054.0520
130.104.0520
130.154.0520
130.254.0520

Product NameiHILIC®-Fusion(P), HILIC Column, PEEK, 20x2.1mm, 5µm, 200ÅiHILIC®-Fusion(P), HILIC Column, PEEK, 30x2.1mm, 5µm, 200ÅiHILIC®-Fusion(P), HILIC Column, PEEK, 50x2.1mm, 5µm, 200ÅiHILIC®-Fusion(P), HILIC Column, PEEK, 100x2.1mm, 5µm, 200ÅiHILIC®-Fusion(P), HILIC Column, PEEK, 150x2.1mm, 5µm, 200ÅiHILIC®-Fusion(P), HILIC Column, PEEK, 250x2.1mm, 5µm, 200ÅiHILIC®-Fusion(P), HILIC Column, PEEK, 50x4.6mm, 5µm, 200ÅiHILIC®-Fusion(P), HILIC Column, PEEK, 100x4.6mm, 5µm, 200ÅiHILIC®-Fusion(P), HILIC Column, PEEK, 150x4.6mm, 5µm, 200ÅiHILIC®-Fusion(P), HILIC Column, PEEK, 250x4.6mm, 5µm, 200Å

価格
¥144,000¥154,000¥166,000¥181,000¥206,000¥245,000¥181,000¥203,000¥236,000¥294,000

P/N
160.022.0520
160.032.0520
160.052.0520
160.102.0520
160.152.0520
160.252.0520
160.054.0520
160.104.0520
160.154.0520
160.254.0520

Product NameiHILIC®-(P) Classic, HILIC Column, PEEK, 20x2.1mm, 5µm, 200ÅiHILIC®-(P) Classic, HILIC Column, PEEK, 30x2.1mm, 5µm, 200ÅiHILIC®-(P) Classic, HILIC Column, PEEK, 50x2.1mm, 5µm, 200ÅiHILIC®-(P) Classic, HILIC Column, PEEK, 100x2.1mm, 5µm, 200ÅiHILIC®-(P) Classic, HILIC Column, PEEK, 150x2.1mm, 5µm, 200ÅiHILIC®-(P) Classic, HILIC Column, PEEK, 250x2.1mm, 5µm, 200ÅiHILIC®-(P) Classic, HILIC Column, PEEK, 50x4.6mm, 5µm, 200ÅiHILIC®-(P) Classic, HILIC Column, PEEK, 100x4.6mm, 5µm, 200ÅiHILIC®-(P) Classic, HILIC Column, PEEK, 150x4.6mm, 5µm, 200ÅiHILIC®-(P) Classic, HILIC Column, PEEK, 250x4.6mm, 5µm, 200Å

価格
¥144,000¥154,000¥166,000¥181,000¥206,000¥245,000¥181,000¥203,000¥236,000¥294,000

131.122.0520
131.322.0520

iHILIC®-Fusion(P), Guard Column, PEEK, 20x2.1mm, 5µm, 200Å, 1pk
iHILIC®-Fusion(P), Guard Column, PEEK, 20x2.1mm, 5µm, 200Å, 3pk*

¥89,000¥221,000

161.122.0520
161.322.0520

iHILIC®-(P) Classic, Guard Column, PEEK, 20x2.1mm, 5µm, 200Å, 1pk
iHILIC®-(P) Classic, Guard Column, PEEK, 20x2.1mm, 5µm, 200Å, 3pk*

¥89,000¥221,000

価格¥172,000¥193,000¥230,000

P/N
164.053.0520
164.103.0520
164.153.0520

価格¥227,000¥242,000¥264,000¥258,000¥267,000¥279,000¥304,000

P/N
180.051.0520
180.101.0520
180.151.0520
180.052.0520
180.102.0520
180.152.0520
180.252.0520

iHILIC®-(P) Classic - HPLC HILIC Column in SS

iHILIC®-Fusion(P) - HPLC HILIC Column in SS
P/N
134.053.0520
134.103.0520
134.153.0520

Product NameiHILIC®-Fusion(P), HILIC Column, SS, 50x3.0mm, 5µm, 200ÅiHILIC®-Fusion(P), HILIC Column, SS, 100x3.0mm, 5µm, 200ÅiHILIC®-Fusion(P), HILIC Column, SS, 150x3.0mm, 5µm, 200Å

Product NameiHILIC®-Fusion(P), HILIC Column, PEEK-SS, 50x1mm, 5µm, 200ÅiHILIC®-Fusion(P), HILIC Column, PEEK-SS, 100x1mm, 5µm, 200ÅiHILIC®-Fusion(P), HILIC Column, PEEK-SS, 150x1mm, 5µm, 200ÅiHILIC®-Fusion(P), HILIC Column, PEEK-SS, 50x2.1mm, 5µm, 200ÅiHILIC®-Fusion(P), HILIC Column, PEEK-SS, 100x2.1mm, 5µm, 200ÅiHILIC®-Fusion(P), HILIC Column, PEEK-SS, 150x2.1mm, 5µm, 200ÅiHILIC®-Fusion(P), HILIC Column, PEEK-SS, 250x2.1mm, 5µm, 200Å

価格
¥172,000¥193,000¥230,000iHILIC®-(P) Classic - HPLC HILIC Column in PEEK-SS

iHILIC®-Fusion(P) - HPLC HILIC Column in PEEK-SS
P/N
170.051.0520
170.101.0520
170.151.0520
170.052.0520
170.102.0520
170.152.0520
170.252.0520

Product NameiHILIC®-(P) Classic, HILIC Column, SS, 50x3.0mm, 5µm, 200ÅiHILIC®-(P) Classic, HILIC Column, SS, 100x3.0mm, 5µm, 200ÅiHILIC®-(P) Classic, HILIC Column, SS, 150x3.0mm, 5µm, 200Å

Product NameiHILIC®-(P) Classic, HILIC Column, PEEK-SS, 50x1mm, 5µm, 200ÅiHILIC®-(P) Classic, HILIC Column, PEEK-SS, 100x1mm, 5µm, 200ÅiHILIC®-(P) Classic, HILIC Column, PEEK-SS, 150x1mm, 5µm, 200ÅiHILIC®-(P) Classic, HILIC Column, PEEK-SS, 50x2.1mm, 5µm, 200ÅiHILIC®-(P) Classic, HILIC Column, PEEK-SS, 100x2.1mm, 5µm, 200ÅiHILIC®-(P) Classic, HILIC Column, PEEK-SS, 150x2.1mm, 5µm, 200ÅiHILIC®-(P) Classic, HILIC Column, PEEK-SS, 250x2.1mm, 5µm, 200Å

価格
¥227,000¥242,000¥264,000¥258,000¥267,000¥279,000¥304,000

PEEK - Poly(ether ether ketone) engineering plastic column hardware
SS - Stainless steel column hardware
PEEK-SS - PEEK-lined stainless steel column hardware* One HPLC column coupler is included

PE - Polyethylence※上記価格は2024年2月1日から適用されます。
※上記価格には消費税は含まれておりません。

輸入元

販売元

株式会社バイオニック

株式会社 クロマニックテクノロジーズ

417-0801 富士市大渕3397-19
TEL: 0545-38-9125 FAX: 0545-38-9126
Email: info@bionikinc.com URL: http:// www.bionikinc.com

552-0001 大阪府大阪市港区波除6-3-1
TEL: 06-6581-0885 FAX: 06-6581-0890
E-mail: info@chromanik.co.jp URL: http://chromanik.co.jp2402