Synos

GENERAL-PURPOSE SYSTEM SOLUTION / COLLIMATED BEAM MEASUREMENT SYSTEMCOLLIMATED BEAM MEASUREMENT SYSTEM
Collimated beam pattern measurement system in combination with collimated beam measurement optics & image processing.

Collimated beam measurement system is for measurement of beam divergence angle ofcollimated beam with high resolution by image processing method. Since the beam divergenceangle of collimated beam can be observed and measured in real time, it can be used for beamquality evaluation of collimator modules and assembling adjustment of collimator lenses, etc.
【Features】○M-Scope type C, collimated beam measurement optics● Quick and easy measurement by dedicated optics and image processing method.
● Divergence angle of collimated beam can be observed and measured in real time with highangular resolution.
○Possible to measure in 400nm to 1700nm wavelength range by selecting detector.
○Optical beam analysis module AP013, specially designed high-functional image processing
software for optical beam proﬁle analysis● All-in-one package of PC, optical beam analysis software, detector driver, correction data.
● High-performance image processing software for optical beam proﬁle measurement Optimetrics BA Standard is pre-installed.
○By combining with various automatic precision positioning stages and image processing, it is possible to automate the collimatedbeam adjustment process for collimator module assembling adjustment.
【Standard component】○Collimated beam measurement optics selection● Focal length 200mm:M-Scope type C/200● Focal length 150mm:M-Scope type C/150● Focal length 100mm:M-Scope type C/100○Available detector selection● 400~1100nm︓Hi-resolution CMOS detector ISA071/ISA071GL● 950~1700nm︓InGaAs NIR detector ISA041H2● 400~1700nm : InGaAs high resolution NIR detector ISA041HRA☞ Regarding the measurement angle coverage and pixel resolution during collimated beam measurement by the detector used, pleaserefer to P50【Detector selection and collimated beam measurement speciﬁcations】○Optical beam analysis module AP013● PC for image processing, optical beam analysis software Optometrics BA Standard, detector driver, calibration data, USB key○Accessories● Cables, instruction manuals, etc.
【Option】○Objective lens
●NFP measurement is also possible if an objective lens is attached. Select the objective lens according to the optical magniﬁcation(ﬁeld of view), pixel resolution, N.A., wavelength, etc.
☞Regarding the ﬁeld of view and pixel resolution during NFP measurement by the detector used, please refer to P51【Detector
selection and NFP measurement speciﬁcations】○ND ﬁlter● Visible (400~700nm): NDF-5 (5 types per set)
● NIR (700~1100nm): NDF NIR-5 (5 types per set)
● IR (1310~1550nm): NDF IR-5（5 types per set）
○Optics bench● Optics bench for ﬁber measurement with manual stages● Vertical setting optics bench【Component selection of collimated beam measurement system】○Stages・optics bench○FFP measurement optics selection○Optical beam analysis module AP013○Detector selection●for visible~1100nm

Sample stages
Optics stages●Personal computer
・Main unit
・Accessories

High resolution CMOS
detector ISA071

Optics bench for ﬁber
measurement●Optical beam analysis software
Optometrics BA Standard●Detector driver●Calibration data●USB licence key●for 950~1700nm

Vertical setting
optics bench

Collimated beam measurement optics
M-Scope type C/200
M-Scope type C/150
M-Scope type C/100

InGaAs high sensitivity NIR
detector ISA041H2

https://www.synos.jp

ND ﬁlter●for 400~1700nm

InGaAs high resolution NIR
detector ISA041HRA/HRVA＊Can be combined with various
motorized/manual stages

21○Accessories

Objective lens

SYNERGY OPTOSYSTEMS CO., LTD