S E L E C T I O N

G U I D E

CATALOG No.5000

Pressure Transmitters
Pressure Transducers

ASAHI GAUGE MFG. CO.,LTD.
J A PA N

Product name

Small-size pressure-resistant &explosion-proof pressure transmitter

Model

1 Ads-MES-Y□

Pressure-resistant & explosion-proof differential pressure transmitter with indicator

2 Ads-MEC-Y□

3 AdC-W-MDC-S□

Pressure-resistant &explosion-proof pressuretransmitter with indicator
4□(2) AdC-MES-S(3) AdC-MEC-S

Appearance

Main unit: f122 x 190

277W x 242H x 140D

277W x 225H x 140D

Application

Authorized product for pressure-resistant &explosion-proof (d2G4)
Mostly for processes• With the field indicator dial
• Authorized product for pressureresistant & explosion-proof
d2G4• With the field indicator dial
• Authorized product for pressureresistant & explosion-proof
d2G4

Pressure range

0~0.1MPa→0~200MPa
Vacuum, compound

0~5kPa→0~0.09MPa
Vacuum, compound

MES-S
0~5kPa→0~0.5MPa
MEC-S

Detection

Wet parts materials

Connection

Power

Output signal
Load resistanceAccuracy (at 20°C)

Model

General type pressuretransmitter
5□(1) MES-C(2) MEC-C

Pressure transmitterwith indicator
6□

MET-B (φ150)
MET-E (φ100)

Small-size pressuretransmitter
7□(1) MES-T249(2) MEC-T249

Small-size pressuretransmitter
8□(1) MES-N249(2) MEC-N249

Appearance

Size

Case

Product nameDrip-proof ADC silver gray

Metal strain gauge

SUS316+SUS630

Drip-proof ADC silver gray
Semiconductor strain
gauge

SUS316+SiO2+NBR

G3/8, 1/2, R3/8, 1/2,
NPT3/8, 1/2

Size

Square 100W x 165H x 60D

Application• General type mainly to beinstalled with equipment• Parallel installation ispossible.

0~0.1MPa→0~200MPa
Vacuum, compound
0~5KPa→0~0.09MPa
Vacuum, compound

Drip-proof, ADC, silver gray

Semiconductor differential pressure
sensor(2) Metal strain gauge(3) Semiconductor pressure sensor

SUS316+SiO2+NBR(2) SUS316+SUS630(3) SUS316+SiO2+NBR

MES-C

Pressure range

0~0.1→0~200MPa
Vacuum, compound

MEC-C 0~20KPa→0~0.09MPa
Vacuum, compound
Drip-proof, ADC, blackCase(1) Metal strain gauge(2) Semiconductor strain gauge

Detection(1) SUS316+SUS630(2) SUS316+SiO2+NBR

Wet parts materials

G, R, NPT3/8, 1/2

G, R, NPT3/8, 1/2

Connection• DC24V ±10%• AC100, 110, 200, 220V ±10%50/60Hz (4-wire)• DC24V ±10%• AC100, 110, 200, 220V ±1050/60Hz (4-wire)• DC24V ±10%• AC100, 110, 200, 220V ±10%50/60Hz (4-wire)Power

DC4~20mA (2,4-wire)
DC0~10mV (3,4-wire)
DC1~5V
(3,4-wire)

DC4~20mA (2,4-wire)
DC0~10mV (4-wire)
DC1~5V(4-wire)DC4~20mA (2,4-wire)
DC0~10mV (3,4-wire)
DC1~5V
(3,4-wire)

Output signal

Output of power current : MAX 550Ω(600Ω for 4-wire)
Output of power voltage : MIN 100kΩ

Output of power current : MAX 550Ω(600Ω for 4-wire)
Output of power voltage : MIN 100kΩ

Output of power current : MAX 550Ω(600Ω for 4-wire)
Output of power voltage : MIN 100k

Load resistance±0.6, 1.0%F·S±1%Indication ±1.6%F.S.
Outputs ±1.0%F.S.

Accuracy (at 20°C)–20~+80°C–10~+40°C

DC4~20mA (2,4-wire)
DC0~10mV (3,4-wire)
DC1~5V
(3,4-wire)
Output of power current : MAX 550Ω(600Ω for 4-wire)
Output of power voltage : MIN 100kΩ±1.0%F·S–10~+60°C–10~+40°C

Others• Mounting on a 2-inch pipe stand or a wall
• Zero adjustment (external adjustment), span
adjustment: ±8%F·S• f50 indicator is also available. (0.1~5MPa)• With a frame arrester• Pressure-resistant packing lead-tool is also available.
• Weight: About 3.2kg.

φ63 x 81

φ42 x 40

• With the field indicator dial

• Compact, light andreasonable price• High accuracy• Applicable for all industries• Compact, light, reasonable
price• High accuracy• Applicable for refrigeratorsand construction machines

0.05MPa→0~50MPa
Vacuum, compound• Zero adjustment, span adjustment:±10%F·S, external adjustment• Weight is about 8.5kg.
• With a frame arrester

1• Zero adjustment ±10%F·S, (external
adjustment)• Weight: About 5.5kg.
• (2) (3) with a frame arrester

Operation temperature of the pressurereceiving part

Others• Self standing or wall-mounting
with 4 screws for installation• Zero adjustment, span
adjustment (all internal
adjustment) ±8%F.S.
• Weight: About 1.2kg

MES

0~0.1MPa→0~600MPa
Vacuum, compoundMES

0~0.1MPa→0~100MPa
Vacuum, compound

MEC

0~5kPa→0~0.09MPa
Vacuum, compound

MEC

0~5kPa→0~0.09MPa
Vacuum, compound

Drip-proof, ADC, black

Drip-proof, ADC, silver

SUS304

Bourdon-tube — Differential
transformer(1) Diaphragm metal strain
gauge(2) Semiconductor strain gauge(1) Diaphragm metal strain
gauge(2) Semiconductor strain gauge

SUS316 or SUS316L(1) SUS304+SUS630(2) SUS304+SiO2+NBR(1) SUS316+SUS630(2) SUS316+SiO2+NBR

G, R, NPT3/8, 1/2

G3/8 (Spec.over 100MPashould be determined by thearrangement.)

G3/8, 7/16 —20UNF

fi DC24V ±10%fi AC100, 110, 200, 220V ±10%50/60Hz (4-wire)fi DC24V ±10%fi DC24V ±10%DC4~20mA (2,4-wire)
DC0~10mV (3,4-wire)
DC1~5V
(3,4-wire)

DC4~20mA (2-wire)
DC1~5V
(3-wire)DC4~20mA (2-wire)
DC1~5V
(3-wire)Output of power current : MAX 550Ω(600Ω for 4-wire)
Output of power voltage : MIN 100kΩ

Output of power current : MAX 550Ω(600Ω for 4-wire)
Output of power voltage : MIN 100kΩ

Output of power current : MAX 550Ω(600Ω for 4-wire)
Output of power voltage : MIN 100kΩ

Indication ±1.6%F.S.
Output ±1.0%F.S.

±0.6%F.S.
±1.0%F.S.

±0.6%FfiS.
±1.0%FfiS.

—5~+45°C—10~+60°C—10~+60°C• Zero adjustment, span
adjustment: ±10%F.S.

• Allowable over pressure:
200%F.S. 20MPa• Vibration resistance: Within
10G• Responsibility: Within 2ms• Weight: About 0.3kg

G1/4, 3/8, R1/4, 3/8,
NPT1/4, 3/8• DC24V ±10%• AC100, 110, 200, 220V ±10%50/60Hz (4-wire)

Operation temperature of the pressurereceiving part

Dial diameter: 100, 150

2• Allowable over pressure:
200%F.S. 20MPa• Vibration resistance: Within
10G• Responsibility: Within 2ms• Weight: About 0.3kg

Product namePressure transmitterfor polystyrene
foam molding

Low range differential pressuretransmitter

Low range differential pressuretransmitter

9 MHC-P□

10 MDC-Y□

11 EMT-1□

Model

Appearance

Size

Application

Product name

Model

Submersible leveltransmitter
12 MES-R□

Sanitary Level And Pressure Transmitters

13 MSC-N□

14 MSC-Y□

Sanitary differential
pressure leveltransmitter
15 MDC-Y□

Appearance

Size

φ300 x About 300H

Main unit: φ44 x 120

Main unit: φ122 x 190

Main unit: φ122 x 190• Measurement of the super low range
differential pressure• High pressure resistant• Measurement of the clean room innerpressure• Low cost

Application• Level detection of the waste
water basin, septic tank, soiltank and dam• Throwing in the liquid for use• Liquid flow arrangement should
be done.

• Level detection of the tankmostly for the food, cosmetics
and pharmaceutical industries• There is no liquid detention.

• Level detection of the tankmostly for the food, cosmetics
and pharmaceutical industries• There is no liquid detention.
• d2G4 explosion proof type isalso available.

• Level detection of closed tanks• Mostly for tanks in the food,cosmetics and pharmaceuticalindustries

Pressure receiving part φ36 x 200 l

Main unit: φ122 x 190

100 x 100 x 110H

• For the polystyrene foam molders• Applicable for high temperature by liquidmedium• Small-size differential pressure transmitter• Draft pressure detection of the stack• Monitoring the filter clogging• Flow rate detection

Pressure range

0.02MPa→0~1MPa

0~10kPa→0~1MPa
Standard pressure less than 1MPa

0~10Pa Pressure resistance
0.1MPa→0~50KPa

Pressure range

0~50kPa (5m)→0~0.5MPa (50m)

0~20KPa→0~1MPa

0~20KPa→0~1MPa

0~15kPa→0~60kPa
Standard pressure MAX.1MPa

Case

Amplifier, ADC material, black

Drip-proof, ADC, silver

Waterproof, ADC, gray

Case

Waterproof, entire exterior SUS

Waterproof, SUS, polished

Waterproof, ADC, silver gray

Drip-proof, ADC, silver gray

Detection

Diaphragm
Semiconductor strain gauge

Semiconductor differential pressure sensor

Diaphragm
Differential transformer

Detection

Metal strain gauge

Semiconductor strain gauge

Semiconductor strain gauge

Semiconductor differential
pressure sensor

Wet parts materials

SUS316+SUS316L (D)

SiO2+SUS316+NBR(or Viton)

Silicon rubber+Aluminum+NBR

Wet parts materials

SUS316+SUS630

SUS316, SUS316L

SUS316, SUS316L

SUS316, SUS316L

Connection

Loose flange connection(Inserting depth is adjustable.)

G3/8, 1/2

Vinyl pipe, rubber tube or metal pipe

Connection

Throw-in type

Flange type, more than JIS10K
50A (Applicable for other various
connections)Flange type, more than JIS10K
50A (Applicable for other various
connections)Flange connection, more than
JIS 10K 50A (Applicable for othervarious connections)Power• DC24V ±10%• AC100, 110, 200, 220V ±10%50/60Hz (4-wire)• DC24V ±10%• AC100, 110, 200, 220V ±10%50/60Hz (4-wire)• DC24V ±10%• AC100, ±10%50/60Hz (4-wire)Power• DC24V ±10%• AC100, 110, 200, 220V ±10%50/60Hz (4-wire)• DC24V ±10%

• DC24V ±10%• AC100, 110, 200, 220V ±10%50/60Hz (4-wire)• DC24V ±10%• AC100, 110, 200, 220V ±10%50/60Hz (4-wire)

Output signal

DC4~20mA (2,4-wire)
DC0~10mV (3,4-wire)
DC1~5V
(3,4-wire)

DC4~20mA (2,4-wire)
DC0~10mV (3,4-wire)
DC1~5V
(3,4-wire)

DC4~20mA (2,4-wire)
DC0~5V
(3,4-wire)
DC1~5V
(3,4-wire)

Output signal

DC4~20mA (2,4-wire)

DC4~20mA (2,4-wire)

DC4~20mA (2,4-wire)
DC0~10mV (3,4-wire)
DC1~5V
(3,4-wire)

DC4~20mA (2,4-wire)
DC0~10mV (3,4-wire)
DC1~5V
(3,4-wire)

Load resistance

Output of power current : MAX 550Ω(600Ω for 4-wire)
Output of power voltage : MIN 100kΩ

Output of power current : MAX 550Ω(600Ω for 4-wire)
Output of power voltage : MIN 100kΩ

500Ω

Load resistance550Ω (600Ω for 4-wire)

MAX : 550Ω (600Ω for 4-wire)

Output of power current : MAX 550Ω(600Ω for 4-wire)
Output of power voltage : MIN 100kΩ

Output of power current : MAX 550Ω(600Ω for 4-wire)
Output of power voltage : MIN 100kΩ

Accuracy (at 20°C)±1.0%F·S±1.0%F·S±1.0%F.S.
(Less than 30Pa: ±2% F·S)

Accuracy (at 20°C)±1.5%F·S±1.0%F·S (–5~40°C)±1.0%F·S (–5~40°C)±1.0%F·S (–5~40°C)

Operation temperature of the pressurereceiving part–20~+150°C–20~+80°C

0~+50°C

Operation temperature of the pressurereceiving part–20~+40°C–10~+150°C–10~+150°C–10~+150°C• Allowable over pressure
200%F.S.
• Liquid medium: Silicon oil• Temperature coefficient:
40Pa/°C(Diaphragm diameter: 72φ)

• Allowable over pressure
200%F.S.
• Liquid medium: Silicon oil• Temperature coefficient:
40Pa/°C(Diaphragm diameter 72φ)
• Mounting on a 2-inch pipe standor a wall for installation• Allowable over pressure
200%F.S.
• Liquid medium: Silicon oil• Temperature coefficient:
40Pa/°C(Diaphragm diameter 72φ)
• Mounting on a 2-inch pipe
stand or a wall for installation• With O-suppression(MAX.120%)• Capillary type MAX.5m

Others• Receiver (analog or digital type) andpressure receiving part and amplifiercompose one set.
• Zero adjustment, span adjustment:±10%F.S.

• Mounting on a 2-inch pipe stand or a wall for
installation• Zero adjustment, span adjustment:±10%F.S.

3• Accessories such as the filter and corrosivegas absorber are available.
• Cable, conduit and metal connector wiring
are available.
• Weight: About 1.1Kg

Others• Using silicon oil for the liquidmedium• SUS or plastic chain is attached.
• Converter is installed outside.
• Weight: About 15.0kg

4Enter each model number.

(Example) M E S - Y 2 0 A
2
4
8

Input

0
4
5

OutputProduct name

Inline type cylinder sealpressure transmitter

Sanitary pressure transmitter

16 MSS-N□□
□

Model

17
□

(1) MEC-T□□2(2) MES-T□□2

Diaphragm seal typepressure transmitter
18 (2) Ads-MES-Y□□2□

Appearance

Product name

Model

(Example) M E S - Y 2 0 A
DC24V (2-wire)
AC100/110V
AC200/220V
DC0~10mV
DC4~20mV
DC1~5V

Types of pressurereceiving part ofdiaphragm

A
B
C
D
E
2

φ8 diaphragm seal
φ10 diaphragm seal
φ15 diaphragm seal
φ18 diaphragm seal
φ24 diaphragm seal
φ28 to φ91 diaphragm seal

Diaphragm Seal Type Pressure Transmitter

Dry type
diaphragm sealpressure transmitter

20 MES-T□□2□
R 1 thread

22 MET-J□□6□

19 MES-J□□□□

21 MES-T□□2□

Appearance

Size

Main unit: φ44 x 120

Cylinder length 154

Main unit: φ122 x 190

Application• Mostly for pipelines in the food,
pharmaceutical and cosmetics
industries• In-line type
• No fluid sticks or residues in thegauge.
• Unnecessary to remove to wash

Pressure range

0~0.1MPa→0~2.5MPa
Vacuum, compound

Case

Size

Dial diameter: φ100, φ150

φ63 x 120

φ63 x 150

Dial diameter: φ100

• Authorized product for pressureresistant & explosion-proof d2G4
• For high temperature, high viscosityand corrosive slurry liquids

Application¥ With the field indicator dial
¥ For high temperature, highviscosity and corrosive slurry
liquids¥ Sanitary type is available.

¥ Compact, light, reasonable
price¥ For high viscosity and fluid
pumping machines¥ Compact, light, reasonable
price¥ For high viscosity and fluid
pumping machines¥ Mostly for the detection of thepressure and high vacuum ofhigh temperature liquid¥ Non-liquid medium type0~0.2MPa→0~10MPa0~0.05MPa→0~10MPa
Vacuum, compound

Pressure range

0~0.05MPa→0~70MPa
Vacuum, compound

0~0.4MPa→0~3.5MPa

0~0.1MPa→0~15MPa—0.1MPa~0
0~0.01MPa→0~1MPa,
including compound type

Waterproof, SUS316, polishedDrip-proof, ADC, silver

Waterproof, ADC, silver

Case

Drip-proof, ADC, black
Drip-proof, SUS304, polishedDrip-proof, ADC, silver

Drip-proof, ADC, silver

Drip-proof, ADC, black

Detection

Diaphragm
Metal strain gauge

Diaphragm(1) Semiconductor strain gauge(2) Metal strain gauge(2) Metal strain gauge

Detection

Bourdon-tube — Metal strain
gauge

Diaphragm metal strain gauge(Front end diaphragm)

Diaphragm
Metal strain gauge

Bellows
Differential transformer

Wet parts materials

SUS316, SUS316L

SUS316, SUS316L

SUS316, SUS316L,
hasteloy, tantalum, titanium, PTFE andothersWet parts materials

SUS316, SUS316L,
hasteloy, tantalum, titanium,
PTFE and others

SUS316, SUS316L

SUS316, 316L,
hasteloy, tantalum, titanium,
PTFE and others

SUS316L+SUS316

Connection

Flange, thread, ferrule and cap nut types

Flange, thread and ferrule types

Flange, thread (G1/2) and protrudent types

Connection

Flange, thread (G1/2) andprotrudent types (Photograph)

Thread type R1BFlange, thread (G1/2) types

Flange, thread types

Power¥ DC24V ±10%

¥ DC24V ±10%

¥ DC24V ±10%¥ AC100, 110, 200, 220V ±10%50/60Hz (4-wire)Power¥ DC24V ±10%¥ AC100, 110, 200, 220V ±10%50/60Hz (4-wire)¥ DC24V ±10%

¥ DC24V ±10%

¥ DC24V ±10%Output signal

DC4~20mA (2,4-wire)

DC4~20mA (2-wire)
DC1~5V
(3-wire)DC4~20mA (2,4-wire)
DC0~10mV (3,4-wire)
DC1~5V
(3,4-wire)

Output signal

DC4~20mA (2,4-wire)
DC0~10mV (3,4-wire)
DC1~5V
(3,4-wire)

DC4~20mA (2-wire)

DC4~20mA (2-wire)
DC1~5V
(3-wire)DC4~20mA (2-wire)
DC1~5V
(3-wire)Load resistance

500Ω

500Ω

500Ω (600Ω for 4-wire)

Load resistance500Ω (600Ω for 4-wire)

MAX : 550Ω

500Ω

500Ω

Accuracy (at 20°C)±1.0%F·S(–5~+40°C)±1.5%F·S±1.5%FÆS

Accuracy (at 20°C)

Output ±1.6%F.S.

±2%FÆS

±1.5%FÆS

±3%FÆS

Operation temperature of the pressurereceiving part–10~+150°C–20~+100°C—50~+200°C(Up to +50°C for vacuum, compound)

Operation temperature of the pressurereceiving part—50~+200°C(Vacuum, compound is up to+50°C)—20~+100°C—50~+200°C(Vacuum, compound is up to+50°C)—20~+450°C• Allowable over pressure: 200%F.S.
• Liquid medium: Silicon oil• Allowable over pressure: 200%F.S.
• Vibration resistance 10G• With zero adjustment, span adjustment±10%F.S.

¥ Mounting on a 2-inch pipe stand or awall for installation¥ Zero adjustment, span adjustment:±10%F.S.
¥ High temperature type for higher than+200°C is available.

Others¥ Zero adjustment, span
adjustment: ±10%F.S.
¥ Capillary type can be
manufactured. (Max.3m)¥ High temperature type forhigher than +200°C isavailable.

¥ Allowable over pressure
200%F.S.
¥ Vibration resistance 10G¥ With zero adjustment, span
adjustment ±10%F.S.

¥ Allowable over pressure
200%F.S.
¥ Vibration resistance 10G¥ With zero adjustment, span
adjustment ±10%F.S.

Others5

6¥ Temperature error is small
due to the non-liquid medium
type¥ Inapplicable for the
coagulating fluid

Enter each model number.

(Example) M E S - Y 2 0 A
2
4
8

Input

0
4
5

OutputSmall-Diaphragm-Seal Pressure Transmitter

Product nameModel

23 MES-Y□□□□

24 MES-C□□□□

25 MES-B□□□□

26 MES-T□□□□

Appearance(Example) M E S - Y 2 0 A
DC24V (2-wire)
AC100/110V
AC200/220V
DC0~10mV
DC4~20mV
DC1~5V

Types of pressurereceiving part ofdiaphragm

Product name

Small-Diaphragm-Seal Pressure Transmitter

Model

27 MGS-C□□□□

28 MES-Z□□□
□

A
B
C
D
E
2

φ8 diaphragm seal
φ10 diaphragm seal
φ15 diaphragm seal
φ18 diaphragm seal
φ24 diaphragm seal
φ28 to φ91 diaphragm seal

Small-Diaphragm-Seal Pressure Transducer
29 TTS-N68□□

30 TTJ-N68□□

Appearance

Size

Main unit φ122 x 190

Main unit: 100W x 130H x 59D

Case diameter: φ150

φ63 x 80

Size

Main unit: 1000W x 130H x 59D

Application• Extruder, plastic molder andadhesive dispenser• For the detection of other hightemperature, high viscosity andcoagulating fluid pressure• Extruder, plastic molder andadhesive dispenser• For the detection of other hightemperature, high viscosityand coagulating fluid pressure• Extruder, plastic molder andadhesive dispenser• For the detection of other hightemperature, high viscosityand coagulating fluid pressure• With the field indication dial

• Extruder, plastic molder andadhesive dispenser• For the detection of other hightemperature, high viscosityand coagulating fluid pressure• Without the field indication dial
• Only non-capillary type

Application• Extruder, plastic molder andadhesive dispenser• For the detection of other hightemperature, high viscosity andcoagulating fluid pressure• With the digital field indicator

Pressure range

0~5MPa→0~70MPa(Varies depending on thediaphragm diameter.)

0~5MPa→0~70MPa(Varies depending on thediaphragm diameter.)

0~5MPa→0~70MPa(Varies depending on thediaphragm diameter.)

0~5MPa→0~70MPa(Varies depending on thediaphragm diameter.)Pressure range

Case

Waterproof, ADC, silver gray

General type, ADC, black

Drip-proof, ADC, black

Drip-proof, ADC, silver

Detection

Diaphragm
Metal strain gauge

Diaphragm
Metal strain gauge

Diaphragm
Metal strain gauge

Wet parts materials

SUS316+SUS316L

SUS316+SUS316L

Connection

G1/4 union, 1/2-20UNF (φ8)
G3/8 union(φ10)
G1/2 union(φ15)
G3/4 union
(φ18)

Power

Main unit: φ38 x 115

Main unit: φ38 x 115• Extruder, plastic molder andadhesive dispenser• For the detection of other hightemperature, high viscosityand coagulating fluid pressure• With the square indicator• Extruder, plastic molder andadhesive dispenser• For the detection of other hightemperature, high viscosityand coagulating fluid pressure• Extruder, plastic molder andadhesive dispenser• For the detection of other hightemperature, high viscosity andcoagulating fluid pressure• Temperature can be detected
simultaneously (with a thermocouple).

0~5MPa→0~70MPa(Varies depending on thediaphragm diameter.)

0~5MPa→0~70MPa(Varies depending on thediaphragm diameter.)

0~10MPa→0~70MPa(Varies depending on thediaphragm diameter.)

0~10MPa→0~70MPa(Varies depending on thediaphragm diameter.)

Case

General type, ADC, black

General type, ADC, black

General type, SUS

General type, SUS

Diaphragm
Metal strain gauge

Detection

Diaphragm
Metal strain gauge

Diaphragm
Metal strain gauge

Diaphragm
Metal strain gauge

Diaphragm
Metal strain gauge

SUS316+SUS316L

SUS316+SUS316L

Wet parts materials

SUS316+SUS316L

SUS316+SUS316L

SUS316+SUS316L

SUS316+SUS316L

G1/4 union, 1/2-20UNF (φ8)
G3/8 union(φ10)
G1/2 union(φ15)
G3/4 union(φ18)

G1/4 union, 1/2-20UNF (φ8)
G3/8 union(φ10)
G1/2 union(φ15)
G3/4 union(φ18)

G1/4 union, 1/2-20UNF (φ8)
G3/8 union(φ10)
G1/2 union(φ15)
G3/4 union(φ18)

Connection

G1/4 union, 1/2-20UNF (φ8)
G3/8 union(φ10)
G1/2 union(φ15)
G3/4 union(φ18)

G1/4 union, 1/2-20UNF (φ8)
G3/8 union(φ10)
G1/2 union(φ15)
G3/4 union(φ18)

G1/4 union
G3/8 union
G1/2 union
G3/4 union

G1/4 union
G3/8 union
G1/2 union
G3/4 union• DC24V ±10%• AC100, 110, 200, 220V ±10%50/60Hz (4-wire)• DC24V ±10%• AC100, 110, 200, 220V ±10%50/60Hz (4-wire)• DC24V ±10%• AC100, 110, 200, 220V ±10%50/60Hz (4-wire)• DC24V ±10%

Power• DC24V ±10%

• DC24V ±10%• AC100, 110, 200, 220V ±10%50/60Hz (4-wire)

Max.loading voltage
DC10V
Option: DC5V

Max.impressed voltage
DC10V
Option: DC5V

Output signal

DC4~20mA (2,4-wire)
DC0~10mV (3,4-wire)
DC1~5V
(3,4-wire)

DC4~20mA (2,4-wire)
DC0~10mV (3,4-wire)
DC1~5V
(3,4-wire)

DC4~20mA (2,4-wire)
DC0~10mV (3,4-wire)
DC1~5V
(3,4-wire)

DC4~20mA (2-wire)
DC1~5V
(3-wire)Output signal

DC4~20mA (2-wire)
DC1~5V
(3-wire)DC4~20mA (2,4-wire)
DC0~10mV (3,4-wire)
DC1~5V
(3,4-wire)

2mV/V and 3.33mV/V
Any output version is available
by making a set with a converter.
Option: 70±25mV

2mV/V and 3.33mV/V
Any output version is available
by making a set with a converter.
Option: 70±25mV

Load resistance500Ω (600Ω for 4-wire)

500Ω (600Ω for 4-wire)

500Ω (600Ω for 4-wire)500Ω

Load resistance500Ω500Ω (600Ω for 4-wire)

Bridge resistance: 350Ω
Option: 3.5±1.5kΩ

Bridge resistance: 350Ω
Option: 3.5±1.5kΩ

Accuracy (at 20°C)±1.0%F·S(Between Dial 20% and 80%)

±1.0%F·S(Between Dial 20% and 80%)

±1.0%F·S(Between Dial 20% and 80%)

±1.0%F·S(Between Dial 20% and 80%)

Accuracy (at 20°C)±1.0%F·S(Between Dial 20% and 80%)

±1.0%F·S(Between Dial 20% and 80%)

±1.0%F·S±1.0%F·S

Operation temperature of–20~+350°C
the pressure receiving part (Mercury) (when pressurized)–20~+350°C(Mercury) (when pressurized)–20~+350°C(Mercury) (when pressurized)–20~+350°C(Mercury) (when pressurized)

Operation temperature of–20~+350°C
the pressure receiving part (Mercury) (when pressurized)–20~+350°C(Mercury) (when pressurized)–20~+350°C(Mercury) (when pressurized)–20~+350°C(Mercury)(when pressurized)• As a liquid medium, φ8, φ10, φ15
uses mercury. Others use
mercury or non-mercuric medium.
• Thermocouple can be mounted.
(Simultaneous measurement of• As a liquid medium, φ8, φ10,temperature and pressure is
φ15 uses mercury.
possible.)
Others use mercury or• Capillary type can be
non-mercuric medium.
manufactured. (Max: φ8 and 3m)• Authorized product for pressureresistant explosion-proof isavailable.

• As a liquid medium, φ8, φ10,
φ15 uses mercury.
Others use mercury or
non-mercuric medium.

• As a liquid medium, φ8, φ10,
φ15 uses mercury.
Others use mercury or
non-mercuric medium.

• As a liquid medium, φ8, φ10,
φ15 uses mercury.
Others use mercury or
non-mercuric medium.
• LED : 3 digits• Sampling: 4 times/sec.

• As a liquid medium, φ8, φ10,
φ15 uses mercury.
• Others use mercury or
non-mercuric medium.

• Capillary type only (Max.2m)• As a pressure medium, φ8, φ10, • Capillary type only (Max.2m)• With the temperature detection
φ15 uses mercury. Others use• Liquid medium is mercury.
mercury or non-mercuric
medium.

Others7

Others

8

(φ8)
(φ10)
(φ15)
(φ18)

(φ8)
(φ10)
(φ15)
(φ18)

Product name

General type transducer

Temperature detection terminal

Sanitary temperaturetransmitter

31 TTS-N
□

32 ——
□

33 ——
□

Model

Appearance

Product name

Digital pressure gauge

34
□

Model

MGS-J
MGC-J

Portable digitalpressure gauge
35□

Digital pressure gauge
with peak hold

MES-ZS
MEC-ZS

36□

Digital pressure gauge
with contacts

MUS-H
MSC-H

37
□

MWS-J
MWC-J

Appearance(Case: A model)Size

24 x 28HEX x 64

Applicationfi Compact, light, reasonable pricefi High accuracy
fi Freezer, construction machines andmedical instruments

Pressure range

0~0.3MPa→0~100MPa
Vacuum, compound

Temperature range
0~— 40°C→0~500°C

Case

Drip-proof, SUS

Main unit: plastic, drip-proof

φ98 x 40D

φ63 x 350

fi Temperature detection of the hightemperature and high pressure fluid atsuch as an extruder
fi Cable end is fitted with the pressuretransmitter of φ6
and φ8.

fi For the fluid temperaturetransmission of food tanks andpipeline

Temperature range
Platinum resistor
element—100↔MAX400°C
Thermocouple MAX1200°C(K)

Size

Dial diameter: f100

Main unit: 92W x 155H x 45D

Main unit: 108W x 130H x 40D

Dial diameter: f100

Application• It can be used instead ofBourdon-tube pressure gauge• Output can be obtained.
• Maximum indication: 0~1999• You can carry the gauge andcheck the pressure at thedesired place.
• It indicates for 20 seconds.
• Power is the battery.
• Maximum indication: 0~1999• Portable to measure thepressure wherever you want
• With function of releasing
inner pressure• ON-OFF control of alarms andmotors and others

Pressure range

0~5kPa→0~100MPa
Compound (same unit), vacuum

0~20kPa→0~70MPa
Compound (same unit),
vacuum0~20kPa→0~35MPa
Compound (same unit),
vacuum

0~5kPa→0~100MPa

Case

Drip-proof, ADC, black

General type, plastic, gray

General type, steel plate

General type, ADC, black

Detection

Diaphragm semiconductor strain
gauge or metal strain gauge

Diaphragm semiconductor
strain gauge or the metal strain
gauge

Diaphragm semiconductor
strain gauge or the metal strain
gauge

Diaphragm semiconductor strain
gauge or the metal strain gauge

Wet parts materials

SUS316+SiO2+NBR
SUS316+SUS630

SUS316 or
BRASS+SiO2+NBR
SUS316+SUS630

SUS316+ Silicon monocrystal+SiO2 or
SUS316+SUS630

SUS316+SiO2+NBR or
SUS316+SUS630

Connection

G, R, NPT3/8, 1/2

G, R, NPT1/4, 3/8,
Hose port (One with thepermanent type is available.)

G, R, NPT1/4, 3/8

G, R, NPT3/8, 1/2

Power• AC100, 110, 200, 220V ±10%50/60Hz (4-wire)
DC24V

U3 dry cell x 4 pieces

AC100, 110V ±10%50/60Hz (4-wire)

AC100V, AC200V, DC24V

Output signal

Option DC4~20mA (4-wire)
DC0~10mV (4-wire)
DC1~5V(4-wire)—Option DC4~20mA

Load resistance

500Ω

—

500Ω

500Ω

Accuracy (at 20°C)±0.5%F·S ±1digit±0.6%F·S ±1digit±1.0%F·S ±1digit±0.6%F·S ±1digit

Operation temperature of the pressurereceiving part–10~+60°C

0~+60°C–10~+60°C–10~+60°C

Others• Case shape: B type andD type are available.
• Indicator: LED 3 digits• Sampling: 2.5 times/sec.
• Contact capacity: AC250V 8A• Diaphragm seal type isavailable.

• Equipped with the automatic
power off function• With the alarm to replacebatteries• Indicator: LDC 3 digits• Sampling: 2.5 times/sec.
• Battery life: About 100 hours• Weight: About 0.4kg• With peak-hold function• BCD parallel output available• Indicator: LED 3 digits• Sampling: 2.5 times/sec.

Drip-proof, ADC, silver

Detection

Diaphragm metal strain gauge

Model K
Thermocouple
Model J

Wet parts materials

SUS316+SUS630

SUS316

Temperature measuring resistor
or thermocouple

SUS316

Connection

G3/8 (Spec.over 100MPa should bedetermined by the arrangement.)

Power

Recommendable impressed voltage: 5V

G1/4 union (φ8)
G3/4 union (φ18)

Ferrule, cap nut and thread
types
—
—

Output signal

5MPa≧60±25mVDC
100MPa≧70±25mVDC

Load resistanceBridge resistance: 3.5±1.5kΩ

Standard output of thermocouple(Example) by JIS
Thermocouple: K1000°C 41.31mV
Pt100Ω 400°C—(4-wire)Option DC4~20mA
DC0~10mV
DC1~5V—

Accuracy (at 20°C)±0.5, 1.0%F·S

0.4CLASS
Within ±0.6%F.S.

Operation temperature of the pressurereceiving part

Others–20~+60°C

Temperature range—• Cable end is open free.
• Specify the cable length.

fi Capillary 2,000~5,000mm

910• Diaphragm type is available.

Product name

Digital differential pressuregauge
38 MDC-J448
□

Model

Pressure-resistant & explosion-proof pressure transmitter with digital indicator39□(1) Adc-MGC-S(2) Adc-MGS-C

Electronic switch

Product name

40 MWT-C□(1 point~3 points)

Model

Receiver42 Meter relay□

43 Digital panel meter□

NRC

DT-5/DM-5

41 Wide-angle receiver□DM-5 (with contacts)

Appearance

Appearance
DT-5(Picture shows the 3-point model.)Size

Dial diameter: f100

277W x 245H x 140D

Main unit: 100W x 130H x 59D

Application• Differential pressure is indicated indigital• With the field indicating dial (digital)• Authorized product for pressureresistant and explosion-proof• Contacts life is long.
• Resistant to the vibration and shocks• Compact and light with 3-point

Pressure range

0~10kPa→0~0.3MPa
Standard pressure is less than 1MPa.

(1) 0~5KPa→0~0.09MPa(2) 0~0.1→0~200MPa0~0.05MPa→0~50MPa

Size

110M x 75D

120W x 100H x 86D

96W x 48H x 145D

Application• Easy view gauge by the wide-angle
dial• Dial with the desired unit is available.

• With upper and lower limits contacts• Large contacts capacity• Dial with the desired unit is available.

• Thermocouple or temperature
measuring resistor input can beindicated.
• LED allows easy reading.
• Large figure with the height of 12.7

Case

Drip-proof, ADC, black

Drip-proof, ADC, silver gray

General type, ADC

Detection

Semiconductor differential pressure
sensor(1) Semiconductor pressure sensor(2) Metal strain gauge

Bourdon-tube or the differential
transformer or the semiconductor straingauge

Wet parts materials

SUS316+SUS316L+NBR(1) SUS316+SiO2+NBR(2) SUS316+SUS630

SUS316 or brass or silicon single crystal+SiO2 + NBR

Input signal

DC4~20mA
DC0~10mV
DC1~5V or others

DC4~20mA
DC0~10mV
DC1~5V or others

DC4~20mA, DC0~10mV, DC1~5V andothers(Transducer output can be directly input
and indicated.)

Connection

G3/8, 1/2

G, R, NPT3/8, 1/2

G, R, NPT1/4, 3/8

Case

General type, plastic

Panel embedded type, plastic

Panel embedded type, plastic, gray

Power

AC100, 110, 200, 220V ±10%50/60Hz DC24V

DC24V ±10%AC100, 110, 200, 220V ±10%50/60Hz (4-wire)

AC100, 110, 200, 220V ±10%50/60Hz

Detection

Moving-coil typeMoving-coil type
Photoelectric non-contact 2ab

Double integral typeContacts analog comparison type 2ab

Output signal

OptionDC4~20mA (4-wire)

DC4~20mA (4-wire)
DC0~10mV (4-wire)
DC1~5V(4-wire)Relay output resistance load AC100V 3A

Power

AC100V or AC200V is required whenthe measurement range is DC0~10mV.

AC100, 110, 200, 220V ±10%50/60Hz• AC100, 110, 200, 220V ±10%50/60Hz

Load resistance500Ω

Output of power current : MAX 550Ω(600Ω for 4-wire)
Output of power voltage : MIN 100kΩ

Accuracy (at 20°C)±1.6%F·S ±1digit±1.0%F·S ±1digit

Contact accuracy 2%F.S.

—

Relay output (H, L)
AC250V, 3A (Resistance load)
DC30V, 3A

Relay output (H, L)
AC250V, 1A (Resistance load)

Operation temperature of the pressurereceiving part–10~+60°C–10~+40°C–10~+60°C

Others• Indicator: LED 3 digits• One side pressure resistance: 0.1MPa• Inapplicable for fluids• Zero adjustment, span adjustment±10%F.S., external adjustment• Setting with the rotary switch• Liquid medium type is available.
• With a frame arrester

11—• Setting by the external volume knob or
the internal rotary switch (2-digit)• Liquid medium type is available.

Output signal

Accuracy (at 20°C)±0.6, 1.0, 1.6%F·S±1.6%F·S±0.2%F·S ±1digit

Allowable ambient
temperature

50°C

50°C

50°C

OthersGraduation unit of dial is MPa, %,NM3/H or others are available on thecustomer’s order.

Graduation unit of dial is MPa, %, NM3/H
or others are available on the customer’sorder.

• Scaling meter: One point fixed(Adjusted by the manufacturer)• 2-wire process digital meter• BCD indicator• Various digital meter relays areavailable.

12Peripheral equipment
44

Power Box Model J006
Rated output voltage: 24V
Raged output current: 0.7A
Input voltage: AC85~264V
Frequency: 47~440Hz
Ripple noise: 150mV MAY
Overcurrent protection: Operation at 105%
Overvoltage protection: 115~140%
In the case of computers and recorders,
use the following stabilized power supply.

45

Distributor Model DS-24
Exclusive for 2-wire type single loop
Stabilized power supply
Rated output: DC24VPower supply: AC100V ±10%46Converter

MAC-C(For diffusion-type semiconductor sensors)

MAS-C(For strain gauges)

The converter is connected with the pressure transducer to convert the pressuretransducer supply power and its output
signal to the output of DC4~20mA.
Power supply
DC24V ±10% (2-wire, 4-wire)
AC100, 110, 200, 220V ±10% 50/60Hz
Dimensions: 108W x 130H x 59D

These specifications are subject to change without notice for improvement.