SB series

SPEC. SHEETRTD Transmitter

Model:SBRFeatures
Simple wiring using a plug-in socket
Compact
3-port insulation (Input-Output-Power)

Model
SBR --Input
01: Pt100
02: JPt100
03: Pt1000
04: CU50
05: CU100
Output
01: 4 to 20mA DC
02: 0 to 20mA DC
03: 0 to 12mA DC
04: 0 to 10mA DC

05: 0 to 1V DC
06: 0 to 5V DC
07: 1 to 5V DC
08: 0 to 10V DC

Power supply
0: 100 to 240V AC
1: 24V DC

How to orderSpecify a model and input range.
(e.g.) SBR-0101-00 to 400

Input specifications
Input detection current: Approx. 0.2mA
Allowable lead wire resistance: 10 or less per wire
Burnout: Upscale, Downscale (Selectable by the DIP switch)
Input:RTD
Input range
Pt100-200 to 850-328 to 1562
JPt100-200 to 500-328 to 932
Pt1000-200 to 850-328 to 1562
CU50-50 to 150-58 to 302
CU100-50 to 150-58 to 302
Minimum span: 50 (100 )Output specifications
When the output range lower limit is zero, (even if zeroadjustment results in a negative value), the output value
will not be negative.
DC current
Outputrange
4 to 20mA DC
0 to 20mA DC
0 to 12mA DC
0 to 10mA DC

AllowableZeroload
adjustment
resistancerange
600 or less -2.5 to 2.5%600 or less0 to 2.5%
1k or less0 to 2.5%
1k or less0 to 2.5%

Spanadjustment
range
97.5 to 102.5%
97.5 to 102.5%
97.5 to 102.5%
97.5 to 102.5%

AllowableZeroload
adjustment
resistancerange
100 or more 0 to 2.5%500 or more 0 to 2.5%500 or more -2.5 to 2.5%
1k or more
0 to 2.5%

Spanadjustment
range
97.5 to 102.5%
97.5 to 102.5%
97.5 to 102.5%
97.5 to 102.5%

DC voltage
Outputrange
0 to 1V DC
0 to 5V DC
1 to 5V DC
0 to 10V DC

PerformanceAccuracy
Response time: Within 0.2%
: 1 sec. (090%)(Average 0.5sec)
Temperature coefficient: 0.015%/
Insulation resistance : 10M or more, at 500V DC(Input-Output-Power)Dielectric strength: 2.0kV AC for 1 minute(Input-Output-Power)

SBR
Instrument specifications
Case: Flame resistant resin Color: Light gray
Front panel: Polycarbonate
Spring type plug: Polyamide Color: Green
Adjustment: By the front potentiometer
Zero adjustment : 2.5%
Span adjustment : 2.5%
Indication:
Power indicator (PWR): Green
Lights when the power is turned on.
Flashes in 500ms cycles when an error has occurredin non-volatile IC memory.
Error indicator (ERR): Red
Flashes in 250ms cycles when input is 110% or more.
Flashes in 500ms cycles when input is -10% or less.
Output status selection: Selects Normal or Reverse with
the DIP switch.
No.1 OFF: Normal, ON: Reverse
Burnout: Selects Upscale or Downscale with the DIPswitch.
No.2 OFF: Upscale, ON: Downscale
Momentary power failure: 30msec.
Self diagnosis:
The CPU is monitored by a watchdog timer, and whenan abnormal status is found on the CPU, the unit
restarts with the reset action.

Ambient temperature : -5 to 55
Ambient humidity: 35 to 85%RH (non-condensing)
Weight: Approx. 80g
Mounting method: DIN rail mounting
Be sure to use fastening plates atboth ends of the unit after the unit
is mounted to the DIN rail.
External dimensions : 17.5 (W) x 75 (H) x 85 (D) mm

Environmental specification
RoHS directive complianceFerrules
Made by Phoenix Contact GMBH & CO.
Insulation sleeve attached
ModelConductor cross section
2AI0.25-6BU
0.2 – 0.25mm
2AI0.34-8TQ
0.25 – 0.34mm
2AI0.5-8WH
0.34 – 0.5mm
2AI0.75-8GY
0.5 – 0.75mm
2AI1-8RD
0.75 – 1.0mm
2AI1.5-8BK
1.0 – 1.5mm
2AI2.5-8BU
1.5 – 2.5mm
Crimping pliers
CRIMPFOX ZA3CRIMPFOX UD6Power indicator
Error indicator

Zero adjustment

Circuit configuration, terminal arrangement

Span adjustment

Power supply
Output+7

Power supply++-6+5-4Powercircuit

OutputcircuitOutputA
Input B
BA

3B

2B

1InputInput
circuitCPU

External dimensions (Scale: mm)
DIP switch
Spring type plug

DIN rail

Installation specifications
Power supply• 100 to 240V AC 50/60Hz
Allowable voltage range: 85 to 264V AC
Power consumption: Approx. 3.5VA• 24V DC
Allowable voltage range: 20 to 28V DC
Power consumption: Approx. 3.5W

SHINKO TECHNOS CO., LTD.

Tel: +81-72-727-6100
URL: http://www.shinko-technos.co.jp/e/E-mail: overseas@shinko-technos.co.jp

20100624