FreeStyle® Comfort®
Portable Oxygen Concentrator with autoSAT®, UltraSense® and autoDOSE®

User Manual (English)

Portable Oxygen Concentrator with UltraSense® and autoDOSE™

Manuel d’utilisation (Français)Concentrateur d’oxygène portable doté d’UltraSense® et d’autoDOSE™

Bruksanvisning (Norsk)Bærbar oksygenkonsentrator med UltraSense® og autoDOSE™
Bruksanvisning (Svenska)

Bärbar syrgaskoncentrator med UltraSense® och autoDOSE™

Bedienungsanleitung (Deutsch)

Käyttöopas (Suomi)

Manual del usuario (Español)Εγχειρίδιο χρήστη (Ελληνικά)Tragbarer Sauerstoffkonzentrator mit UltraSense® und autoDOSE™
Concentrador de oxígeno portátil con UltraSense® y autoDOSE™

Manuale dell’utente (Italiano)

Concentratore di ossigeno portatile con UltraSense® e autoDOSE™

Manual do utilizador (Português)Concentrador de oxigénio portátil com UltraSense® e autoDOSE™

Gebruikershandleiding (Nederlands)
Draagbare zuurstofconcentrator met UltraSense® en autoDOSE™
Brugermanual (Dansk)

Bærbar iltkoncentrator med UltraSense® og autoDOSE™

Kannettava happirikastin UltraSense®- ja autoDOSE™-toiminnoillaΦορητός συμπυκνωτής οξυγόνου με UltraSense® και autoDOSE™

Kullanıcı El Kitabı (Türkçe)
UltraSense® ve autoDOSE™ Özellikli Taşınabilir Oksijen Yoğunlaştırıcı
Uživatelská příručka (Čeština)Přenosný koncentrát kyslíku s UltraSense® a autoDOSE™
Instrukcja obsługi (Polski)

Przenośny koncentrator tlenu z technologią UltraSense® i autoDOSE™

Felhasználói kézikönyv (Magyar)

Hordozható oxigénkoncentrátor UltraSense® és autoDOSE™ segítségével

FreeStyle Comfort
User Controls & System Status Indicators
ISO 7000
Operator’s manual; operating instructions. Reg. # 1641
The country and date of manufacture.
The “CC” identifies the two-lettercountry code of the country of manufacture. The date of manufacture is in
YYYY-MM-DD format. Reg. # 6049

ISO 7010: Graphical symbols—Safety colors andsafety signs—Registered safety signs
The instruction manual must be read.
Reg. # M002
Keep away from open flame, fire,
sparks. Open ignition source andsmoking prohibited. Reg. # P003

Keep away from rain, keep dry. Reg.
# 0626

Do not smoke near unit or while
operating unit. Reg. # P002

Stacking limit by number, where “n”
indicates the maximum number ofunits allowed. Reg. # 2403

Warning. Reg. # W001
IEC 60417
Type BF applied part (degree ofprotection against electric shock).
Reg. # 5333

Catalog Number. Reg. # 2493
Serial Number. Reg. # 2498
Storage or operating temperature
range. Reg. # 0632
Storage humidity range. Reg. # 2620
Atmospheric pressure limitation. Reg.
# 2621Name and address of manufacture.
Reg. # 3082

Class II Equipment, Double InsulatedReg. # 5172
EU Regulation 2017/745
Authorized representative in theEuropean Community
If the product unique device identifier(UDI) label has the CE#### symbolon it, the device complies with therequirements of concerning medical
devices. The CE#### symbol indicates notified body number.
Additional Symbols

This way up. Reg. # 0623

Keep away from flammable materials,
oil and grease.

Fragile, handle with care. Reg. # 0621

Do not disassemble.

Contains hazardous substances. Reg.
# 3723

Increase flow.
Decrease flow.

Importer. Reg. # 3725

Menu button.

O2 Outlet

Power switch. Reg. # 5009
RTCA/DO160 Section
21
Category M
Compliant

2 - EN PN MN232-C4 L | User Manual

The manufacturer of this POC hasdetermined this device conforms to
all applicable FAA acceptance criteria
and use on board aircraft.

FreeStyle Comfort
The manufacturer of this POC hasdetermined this device conforms to
all applicable FAA acceptancecriteria and use on board aircraft.

Council Directive 2012/19/EU
WEEE
This symbol is to remind the equipment owners to return it to a recycling
facility at the end of its life, per Waste
Electrical and Electronic Equipment(WEEE) Directive.
Our products will comply with therestriction of Hazardous Substances(RoHS) directive. They will not contain more than trace amounts of leador other hazardous material content.

Recycle Symbol
Authorized representative in Switzerland (ADD)
If the device bears the UKCA markas shown with UKCA#### indicating
the notified body number, this device
complies with UKCA regulations.
21 CFR 801.15
Federal law restricts this device tosale by or on the order of a physician.
ETL Certification

IEC 60601-1-8
Low Priority Technical Alarm
ISO 15223-1

Certified for both the U.S. and Canadian markets, to the applicable U.S.
and Canadian standards
Contains FCC ID: WAP2001
FCC NoticeThis device may contain CYBLE-022001-00,
including the antenna 2450AT18B100 fromJohnson Technology, complies with par 15 of the
FCC rules. The device meets the requirements
for modular transmitter approved as detailedin FCC public notice DA00-1407. Transmitter
operation is subject to the following two
conditions (1) The device may not cause harmfulinterference, and (2) This device must accept
any interference received, including interference
that may cause undesired operation. The productmay be covered by one or more patents, US andinternational. Please visit our website below
for the listing of applicable patents. Pat: www.
caireinc.com/corporate/patents

Medical Device. Reg. # 5.7.7
Unique device identifier # 5.7.10
A S/NZS 4417.1, AS/NZS 4417.2
ACMA Regulatory ComplianceMark– indicates compliance with
ACMA regulations (ADD)
Wireless Technology
Connection types

SSP, iAP2, GATT

Frequency

2402 to 2480 MHz

Max RF power output+4 dBmOperating range

10m (Class 2)

It is recommended that the machine is at a minimum
distance of 0.43” (1.1 cm) from the body during
operation. Not applicable to masks, air tubing, oraccessories.

Graphical Symbols for Use on Equipment
IEC 60417
Direct Current Reg. # 5031
Alternating Current Reg. # 5032
IEC 60601-1
Drip ProofPN MN232-C4 L | User Manual EN - 3

FreeStyle Comfort
Unpacking Your FreeStyle Comfort
Verify that all of the components listed are included in
the package. If any items are
missing, contact your oxygen
provider immediately.

Standard Accessories• Portable Oxygen Concentrator• Carrying Bag• Shoulder Strap• 8-Cell Battery Pack
• AC Power Supply• AC Power Cord
• DC Power Supply• User Manual

Optional Accessories• Additional 8-Cell Battery Pack• 16-Cell Battery Pack
• Desktop Charger• Carry-all Accessory Bag

Getting to Know Your FreeStyle Comfort
Become familiar with the key features of the FreeStyle Comfort and its user interface.

Front View
Strap Attachment

User Interface

OxygenOutletAir Outlet

Air Intake/Filters
Air Intake/
Filters
Power Port
Receptacle
Tab
Battery Pack
Battery Pack
Release Button
4 - EN PN MN232-C4 L | User ManualFreeStyle Comfort
Top View
Display Screen
Increase Flow

Power Button

Menu Button

Decrease Flow

Display Screen On Battery
Flow Setting

Battery Display

autoSAT Indicator
Battery
Percentage
Remaining

Alarm Display
Bolus Delivery
Indicator

Display Screen On AC/DC Power
Flow Setting
Battery Display

Alarm Display
Space

Charge Indicator

PN MN232-C4 L | User Manual EN - 5

FreeStyle Comfort
What is the Oxygen Concentrator
The air we breathe contains approximately 21%
oxygen, 78% nitrogen, and 1% other gasses. In
the FreeStyle® Comfort®, room air is drawn into
the machine through the air intakes. It then passes
through an adsorbent material called molecularsieve. This material separates the oxygen fromthe nitrogen and allows only the oxygen to passthrough. The result is a flow of high-concentration
oxygen delivered to the user.
The FreeStyle Comfort combines advanced oxygen
concentrator and oxygen conserving technologies
to create a portable device that allows for maximum
portability and ambulation. The advanced pulse
flow delivery quickly senses when the user breathes
in and delivers a pulse of oxygen at the beginningof each inhalation. Pulse flow delivery is ideal for
activities and time away from home. It conserves
oxygen and extends battery life as opposed to continuous flow. The FreeStyle Comfort is lightweight
enough to be carried via shoulder strap. It can be
operated on AC (wall), DC (vehicle) or battery
power.

Intended Use
The CAIRE FreeStyle Comfort Oxygen Concentrator is intended for the administration of supplemental oxygen. The device is not intended for life
support nor does it provide any patient monitoring
capabilities.

Why Your Physician Prescribed
OxygenMany people suffer from a variety of heart, lung,and other respiratory diseases. A significant numberof these people can benefit from supplemental
oxygen therapy at home, when traveling, or while
participating in daily activities away from home.
Oxygen is a gas that makes up 21% of the ambient
air we breathe, and our bodies depend on a steady

6 - EN PN MN232-C4 L | User Manualsupply to function properly. Your physician prescribed a flow or setting to address your particularrespiratory condition.
Although oxygen is a non-addictive drug, unauthorized oxygen therapy can be dangerous. You mustseek medical advice before you use this oxygen
concentrator. The Equipment Provider who supplies
your oxygen equipment will demonstrate how to set
the prescribed flow rate.

Operator Profile
Concentrators are intended to supply supplemental
oxygen to users suffering from discomfort due toailments which effect the efficiency of one’s lungs
to transfer oxygen in the air to their bloodstream.
Portable oxygen concentrators (POCs) do not store
or contain oxygen. They do not need to be refilled,
and can recharge anywhere AC or DC power isavailable (except on an airplane). This makes theuser relatively self-sufficient in terms of in-homeuse, ambulation (both within and outside of thehome), mobility, and overall lifestyle. Oxygen concentrator use requires a physician’s prescription.
The typical oxygen therapy user is older than 65
years of age and suffers from a variety of respiratory diseases, including Chronic Obstructive Pulmonary Disease (COPD). Users typically have good
cognitive abilities and must be able to communicate
discomfort. If the user is unable to communicate
discomfort, or unable to read and understand theconcentrator labeling and instructions for use, thenuse is recommended only under the supervision of
one who can. If any discomfort is felt while using
the concentrator, users are advised to contact theirhealthcare provider. Users are also advised to haveback-up oxygen available (i.e. cylinder oxygen) in
the event of a power outage or concentrator failure.
There are no other unique skills or user abilities
required for concentrator use.

FreeStyle Comfort
Important!
Safety Instructions are defined as follows:
WARNING: IMPORTANT SAFETYINFORMATION FOR HAZARDS THATMIGHT CAUSE SERIOUS INJURY.
CAUTION: Important information forpreventing damage to the FreeStyle
Comfort.
Note: Information needing special attention.

Indications for Use
WARNING: IT IS VERY IMPORTANT
TO SELECT ONLY THE PRESCRIBEDLEVEL OF OXYGEN. DO NOT CHANGE
THE FLOW SELECTION UNLESS YOUHAVE BEEN DIRECTED TO DO SO BY A
LICENSED CLINICIAN.
WARNING: THIS DEVICE HAS NOT BEEN TESTEDOR APPROVED FOR USE WITH PEDIATRICPATIENTS.
WARNING: THIS DEVICE HAS NOT BEEN TESTEDOR APPROVED FOR USE WITH PEDIATRICPATIENTS.
WARNING: FEDERAL (USA) LAW RESTRICTS THIS
DEVICE TO SALE OR RENTAL BY ORDER OF APHYSICIAN OR OTHER LICENSED HEALTH CAREPROVIDER.
WARNING: THIS UNIT IS NOT TO BE USED FOR
LIFE SUPPORT. ADDITIONAL MONITORING MAY
BE REQUIRED FOR ANY USER WHO IS UNABLE
TO COMMUNICATE DISCOMFORT WHILE USING
THIS DEVICE. USERS WITH HEARING AND/OR
SIGHT IMPAIRMENT(S) MAY NEED ASSISTANCE
WITH MONITORING ALARMS.
WARNING: PULSE DOSE MODE SETTINGS MUSTBE DETERMINED FOR EACH USER INDIVIDUALLYFOR THEIR NEEDS AT REST, DURING EXERCISE,
AND WHEN TRAVELING.

WARNING: SOME RESPIRATORY EFFORTS OF
THE PATIENT MIGHT NOT TRIGGER THE CONSERVING FUNCTION.
WARNING: THE FREESTYLE COMFORT IS NOTINTENDED FOR USE WITH A TRACHEOTOMISEDPATIENT.
WARNING: PREGNANT OR NURSING WOMENSHOULD NOT USE ACCESSORIES RECOMMENDED IN THIS MANUAL, THEY MAY CONTAINPHTHALATES.

Contraindications for Use
WARNING: IN CERTAIN CIRCUMSTANCES, THE USE OF NON-PRESCRIBEDOXYGEN CAN BE HAZARDOUS. THIS
DEVICE SHOULD ONLY BE USED WHEN
PRESCRIBED BY A PHYSICIAN.
WARNING: THIS DEVICE IS NOT INTENDED FOR
USE IN OR NEAR MAGNETIC RESONANCE ENVIRONMENTS, SUCH AS MRI MACHINES.
WARNING: NOT FOR USE IN THE PRESENCE OF
FLAMMABLE ANESTHETICS.
WARNING: AS WITH ANY ELECTRICALLYPOWERED DEVICE, THE USER MAY EXPERIENCEPERIODS OF NON-OPERATION AS A RESULT OF
ELECTRICAL POWER INTERRUPTION, OR THE
NEED TO HAVE THE OXYGEN CONCENTRATORSERVICED BY A QUALIFIED TECHNICIAN. THE
OXYGEN CONCENTRATOR IS NOT APPROPRIATE FOR ANY USER WHO WOULD EXPERIENCEADVERSE HEALTH CONSEQUENCES AS THE
RESULT OF SUCH TEMPORARY INTERRUPTION.
WARNING: THIS DEVICE IS NOT APPROVED FOR
USE IN ALL ENVIRONMENTS. EXPOSURE TO
CERTAIN ELECTROMAGNETIC SOURCES MAY
RESULT IN INTERFERENCE OR UNINTENDED
DEVICE PERFORMANCE. IN THE EVENT OF
INTERFERENCE THE DEVICE SHOULD BE SHUT
DOWN IMMEDIATELY.

WARNING: THIS DEVICE IS INTENDED FOR USE IN
THE HOME HEALTHCARE AND SIMILAR OUTPATIENT ENVIRONMENTS.
WARNING: THIS DEVICE IS INTENDED TO BE
USED BY THE PATIENT RECIEVING OXYGENTHERAPY.

PN MN232-C4 L | User Manual EN - 7

FreeStyle Comfort
Safety Guidelines
WARNING: NO MODIFICATION OF THISEQUIPMENT IS PERMITTEDWARNING: THIS DEVICE SUPPLIES HIGH-CONCENTRATION OXYGEN THAT PROMOTES RAPID
BURNING. DO NOT ALLOW SMOKING OR OPEN
FLAMES WITHIN THE SAME ROOM OF (1) THIS
DEVICE, OR (2) ANY OXYGEN-CARRYING ACCESSORY. FAILURE TO OBSERVE THIS WARNING CANRESULT IN SEVERE FIRE, PROPERTY DAMAGE,
AND/OR CAUSE PHYSICAL INJURY OR DEATH.
WARNING: DO NOT OPERATE UNIT IN A RESTRICTED OR CONFINED SPACE (I.E., A SMALL
CASE OR HANDBAG) WHERE VENTILATION CAN
BE LIMITED. THIS CAN CAUSE THE OXYGENCONCENTRATOR TO OVERHEAT AND IMPAIRPERFORMANCE.
WARNING: THE CONCENTRATOR SHOULD BE
LOCATED AS TO AVOID SMOKE, POLLUTANTS ORFUMES.
WARNING: IN THE EVENT THERE IS A SERIOUS
INCIDENT OCCURRING WITH THIS DEVICE, THE
USER SHOULD IMMEDIATELY REPORT THE
INCIDENT TO THE PROVIDER AND/OR THE MANUFACTURER. A SERIOUS INCIDENT IS DEFINEDAS AN INJURY, DEATH, OR POTENTIAL TO CAUSE
INJURY/DEATH SHOULD THERE BE A REOCCURRENCE OF THE INCIDENT. THE USER CAN ALSOREPORT THE INCIDENT TO THE COMPETENT AUTHORITY IN THE COUNTRY WHERE THE INCIDENTOCCURRED.
WARNING: THE USE OF SOME OXYGENADMINISTRATING ACCESSORIESNOT SPECIFIED FOR THIS OXYGENCONCENTRATOR MAY IMPAIR ITSPERFORMANCE. RECOMMENDED ACCESSORIES ARE REFERENCED WITHIN
THIS MANUAL.
WARNING: IF THE OXYGEN CONCENTRATOR HAS
BEEN DROPPED, DAMAGED OR EXPOSED TO
WATER, PLEASE CONTACT YOUR EQUIPMENTPROVIDER FOR INSPECTION OR POSSIBLE REPAIR OF THE DEVICE. DO NOT USE THE OXYGENCONCENTRATOR IF IT HAS A DAMAGED POWER
CORD OR PLUG.
WARNING: PROPERLY SECURE, BELT OR OTHERWISE RESTRAIN THE OXYGEN CONCENTRATOR
WHEN IN A VEHICLE DURING TRANSPORT TO
PREVENT DAMAGE OR INJURY.

8 - EN PN MN232-C4 L | User Manual

WARNING: THE US DEPARTMENT OF TRANSPORTATION (DOT) AND UNITED NATIONS (UN)REGULATIONS REQUIRE THE REMOVAL OF THE
BATTERY FROM THE DEVICE FOR ALL INTERNATIONAL AIRLINE TRAVEL WHEN THE OXYGENCONCENTRATOR IS CHECKED AS LUGGAGE.
WHEN SHIPPING THE OXYGEN CONCENTRATOR,
THE BATTERY MUST ALSO BE REMOVED FROM
THE DEVICE AND PACKAGED PROPERLY.
WARNING: DO NOT OPERATE THIS EQUIPMENTWITHOUT FIRST READING AND UNDERSTANDING
THIS MANUAL. IF YOU ARE UNABLE TO UNDERSTAND THE WARNINGS AND INSTRUCTIONS,CONTACT YOUR EQUIPMENT PROVIDER BEFOREATTEMPTING TO USE THIS EQUIPMENT; OTHERWISE, INJURY OR DAMAGE MAY RESULT.
WARNING: IF YOU FEEL DISCOMFORT OR ARE
EXPERIENCING A MEDICAL EMERGENCY, SEEKMEDICAL ASSISTANCE IMMEDIATELY.
WARNING: OPERATING THE OXYGEN CONCENTRATOR OUTSIDE OF THE OPERATIONALTEMPERATURE SPECIFICATIONS CAN LIMIT THE
CONCENTRATOR’S ABILITY TO MEET OXYGENCONCENTRATION SPECIFICATION. REFER TO THE
SPECIFICATION SECTION OF THIS MANUAL FORTEMPERATURE LIMITS.
WARNING: USE NO OIL, GREASE, OR PETROLEUM-BASED OR OTHER FLAMMABLE PRODUCTS WITH THE OXYGEN-CARRYING ACCESSORIES OR THE OXYGEN CONCENTRATOR. ONLYWATER BASED, OXYGEN COMPATIBLE LOTIONSOR SALVES SHOULD BE USED. OXYGEN ACCELERATES THE COMBUSTION OF FLAMMABLE
SUBSTANCES.
WARNING: THE INCORRECT USE OF THE BATTERY CAN CAUSE THE BATTERY TO GET HOT,IGNITE, AND MAY CAUSE SERIOUS INJURY. BE
SURE NOT TO PIERCE, STRIKE, STEP ON, DROP
THE BATTERY, OR OTHERWISE SUBJECT THE
BATTERY TO STRONG IMPACTS OR SHOCKS.
THE USE OF A DAMAGED BATTERY MAY CAUSEPERSONAL INJURY.
WARNING: DO NOT SHORT-CIRCUIT THE
BATTERY’S METAL CONTACTS WITH METALLICOBJECTS, SUCH AS KEYS OR COINS. THIS MAY
CAUSE SPARKS OR EXCESSIVE HEAT TO BE
GENERATED.

FreeStyle Comfort
WARNING: WHILE USING THE PORTABLE OXYGEN CONCENTRATOR OUTDOORS WITH THE AC
POWER SUPPLY, CONNECT THE POWER SUPPLYINTO A GROUND FAULT INTERRUPTED (GFI)
OUTLET ONLY.
WARNING: THE OXYGEN CONCENTRATORSHOULD NOT BE USED ADJACENT TO OR
STACKED WITH OTHER EQUIPMENT. IF ADJACENT OR STACKED USE IS UNAVOIDABLE, THE
DEVICE SHOULD BE OBSERVED TO VERIFYNORMAL OPERATION.
WARNING: USE OF CABLES AND ADAPTERS
OTHER THAN THOSE SPECIFIED, WITH THE
EXCEPTION OF CABLES AND ADAPTERS SOLDBY THE MANUFACTURER OF THE MEDICAL ELECTRICAL EQUIPMENT AS REPLACEMENT PARTS
FOR INTERNAL COMPONENTS, MAY RESULT IN
INCREASED EMISSIONS OF DECREASED IMMUNITY OF THE OXYGEN CONCENTRATOR.
WARNING: USE ONLY ELECTRICAL VOLTAGE
SPECIFIED ON THE SPECIFICATION LABELAFFIXED TO THE DEVICE.
WARNING: DO NOT USE EXTENSION CORDS WITH
THIS UNIT OR CONNECT TOO MANY PLUGS INTO
THE SAME ELECTRICAL OUTLET. THE USE OF
EXTENSION CORDS COULD ADVERSELY AFFECT
THE PERFORMANCE OF THE DEVICE. TOO MANY
PLUGS INTO ONE OUTLET CAN RESULT IN AN
OVERLOAD TO THE ELECTRICAL PANEL CAUSING THE BREAKER/FUSE TO ACTIVATE OR FIRE IF
THE BREAKER OR FUSE FAILS TO OPERATE.
CAUTION: In the event of an alarm or
you observe the oxygen concentrator is
not working properly; consult the Troubleshooting section in this manual. If
you cannot resolve the problem, consult
your Equipment Provider.
CAUTION: To prevent a void warranty, follow all
manufacturers’ instructions.

WARNING: DO NOT ATTEMPT ANY
MAINTENANCE OTHER THAN THE
POSSIBLE SOLUTIONS LISTED WITHIN
THIS MANUAL.
WARNING: ONLY ACCESSORIES RECOMMENDED
BY THE MANUFACTURER. USE OF ANY OTHER
MAY BE HAZARDOUS, CAUSE SERIOUS DAMAGE
TO YOUR OXYGEN CONCENTRATOR AND WILL
VOID THE WARRANTY .
WARNING: THE MANUFACTURER RECOMMENDS
AN ALTERNATE SOURCE OF SUPPLEMENTALOXYGEN IN THE EVENT OF A POWER OUTAGE,
ALARM CONDITION, OR MECHANICAL FAILURE.
CONSULT YOUR PHYSICIAN OR EQUIPMENTPROVIDER FOR THE TYPE OF RESERVE SYSTEMREQUIRED.
WARNING: DO NOT ALLOW EITHER THE AIRINTAKE OR THE AIR OUTLET VENTS TO BECOMEBLOCKED. DO NOT DROP OR INSERT ANYOBJECTS INTO ANY OPENINGS ON THE DEVICE.
THIS CAN CAUSE THE OXYGEN CONCENTRATORTO OVERHEAT AND IMPAIR PERFORMANCE.
WARNING: SMOKING WHILE USING OXYGEN IS
THE NUMBER ONE CAUSE OF FIRE INJURIES AND
RELATED DEATHS. YOU MUST FOLLOW THESE
SAFETY WARNINGS:WARNING: DO NOT ALLOW SMOKING, CANDLES,OR OPEN FLAMES IN THE SAME ROOM WITH THE
DEVICE OR THE OXYGEN-CARRYING ACCESSORIES.
WARNING: SMOKING WHILE WEARING AN OXYGEN CANNULA MAY CAUSE FACIAL BURNS AND
POSSIBLY DEATH.
WARNING: IF YOU SMOKE, THESE 3 STEPSMAY SAVE YOUR LIFE: TURN OFF THE OXYGENCONCENTRATOR, TAKE OFF THE CANNULA,
AND LEAVE THE ROOM WHERE THIS DEVICE IS
LOCATED.

PN MN232-C4 L | User Manual EN - 9

FreeStyle Comfort
WARNING: “NO SMOKING – OXYGEN IN USE”
SIGNS MUST BE PROMINENTLY DISPLAYEDIN THE HOME, OR WHERE OXYGEN IS IN USE.
USERS AND THEIR CAREGIVERS MUST BE INFORMED ABOUT THE DANGERS OF SMOKING IN
THE PRESENCE OF, OR WHILE USING, MEDICALOXYGEN.
WARNING: DO NOT USE YOUR OXYGEN CONCENTRATOR IN THE PRESENCE OF FLAMMABLE
GASES. THIS CAN RESULT IN RAPID BURNINGCAUSING PROPERTY DAMAGE, BODILY INJURIESOR DEATH.
WARNING: REMOVING THE CANNULA AND
PUTTING IT ON CLOTHING, BEDDING, SOFAS,OR OTHER CUSHION MATERIAL WILL CAUSE A
FLASH FIRE WHEN EXPOSED TO A CIGARETTE,HEAT SOURCE, OR FLAME.
WARNING: DO NOT LEAVE A NASAL CANNULA
ON OR UNDER CLOTHING, BED COVERINGS OR
CHAIR CUSHIONS. IF THE UNIT IS TURNED ON
BUT NOT IN USE, THE OXYGEN WILL MAKE THE
MATERIAL FLAMMABLE. ENSURE THE POWER
IS OFF WHEN THE OXYGEN CONCENTRATOR IS
NOT IN USE.
WARNING: APPROVED OXYGEN DELIVERYACCESSORIES ARE DESIGNED TO REDUCE THE
PROPAGATION OF FIRE FOR THE SAFETY OF THE
PATIENT. USE OF UNAPPROVED ACCESSORIESMAY INCREASE THE RISK OF INJURY OR DAMAGE RELATED TO FIRES.
WARNING: THE OXYGEN CONCENTRATOR MUSTBE OFF AND DISCONNECTED FROM EXTERNALPOWER BEFORE REMOVING THE BATTERY PACK.
WARNING: SMALL PARTS MAY PRESENT A POTENTIAL CHOKING HAZARD.
WARNING: AVOID CONTACT WITH EXHAUST AIR
VENT FOR LONGER THAN 10 SECONDS AS IT
MAY BE WARM.
WARNING: WIND OR STRONG DRAUGHTS CANADVERSELY AFFECT ACCURATE DELIVERY OF
OXYGEN THERAPY.
WARNING: THIS PRODUCT CAN EXPOSE YOU
TO CHEMICALS INCLUDING NICKEL, WHICH
IS KNOWN TO THE STATE OF CALIFORNIA TO
CAUSE CANCER. FOR MORE INFORMATION, GO
TO WWW.P65WARNINGS.CA.GOV.
CAUTION: Always disconnect AC power
supply from the wall before disconnecting the AC power supply from theoxygen concentrator.
10 - EN PN MN232-C4 L | User Manual

CAUTION: Do not position the oxygen concentrator
so that it is difficult to access the power cord.
CAUTION: Always place oxygen supply tubing andpower cords in a manner that prevents trip hazard
or possible accidental strangulation.
CAUTION: When using the oxygen concentrator in
an automobile, boat, or on other DC sources with
the DC power supply, make sure that the vehicle isstarted and running before connecting the oxygen
concentrator. Failure to follow these instructions
can result in the power supply not supplying power
to the oxygen concentrator.
CAUTION: When the automobile in which you
are using the oxygen concentrator unit is turnedoff, disconnect and remove the device from theautomobile. Do not store the oxygen concentrator
in a very hot or cold automobile or in other similar,
high- or low-temperature environments.
Note: Portable and mobile RF communications equipment can effect medical electrical equipment.
CAUTION: DO NOT leave the oxygen
concentrator or the power supply
plugged into the vehicle if the ignition isin the OFF position. Doing so may drain
the vehicle’s battery.
CAUTION: Operating or storing the oxygen concentrator outside of its normal operating temperature
range and ambient pressure range (altitude) canaffect performance and decrease battery run time
and/or increase battery charge time. Refer to theSpecifications section in this manual for storageand operating temperature limits.
CAUTION: For oxygen concentrators equippedwith batteries: Store in a cool and dry location to
help ensure the longevity of your battery. Storing
your oxygen concentrator for extended periods oftime at high temperatures or with a fully charged
or completely discharged battery can degrade
its overall battery life. Do not attempt to open thebattery; there are no serviceable parts inside thebattery. Keep batteries away from children.
CAUTION: ONLY USE the Manufacturer’s provided
batteries. For proper battery disposal, contact
your Equipment Provider or your local government
agency for disposal requirements.

FreeStyle Comfort
Note: To mitigate harm caused by concentrator or
power failure, equipment provider should assess
the need for back-up oxygen during installation andperiodically when these attributes change:-The condition of the patient-The environment in which the patient lives-The ability to resupply the patient with back-up
supplies of supplementary oxygen.
Note: Ensure the cannula is fully inserted and secure.
This ensures that the oxygen concentrator canproperly detect inspiration for oxygen delivery. During
inhalation, you should hear or feel oxygen flow to theprongs of the nasal cannula.
Note: Lithium batteries may permanently lose capacity
when exposed to extremely hot temperatures with thebatteries fully charged or completely depleted. For
extended storage, it is recommended that batteries becharged 25 to 50% and remain within a temperature
range of -4°F – 68°F (-20°C – 20°C).

Note: If the oxygen concentrator has been stored
for an extended period of time outside of its normal
environmental operating conditions shown in thespecifications table, it should be allowed to return tonormal operating temperature before being poweredon.
Note: Equipment provider should assess the needfor back-up oxygen in case of concentrator or power
failure at installation and periodically when the these
attributes change:-The condition of the patient-The environment in which the patient lives-The ability to resupply the patient with back-up
supplies of supplementary oxygen.

Note: Do not operate the oxygen concentrator without
the air intake filter in place. If a second filter is provided, insert the replacement filter before you clean the
dirty filter. Clean the dirty filter in a warm soap andwater solution, then dry thoroughly prior to use.
Note: The manufacturer does not recommend thesterilization of this equipment.
Note: There is never a danger of depleting the oxygenin a room when you use your oxygen concentrator
unit.
WARNING: THE OXYGEN CONCENTRATOR SHOULD NOT BE OPERATEDOR PLUGGED IN TO EXTERNAL (AC/DC) POWER WITHOUT A BATTERYINSTALLED. DO NOT ATTEMPT TO
POWER ON THE UNIT OR CONNECT
IT TO EXTERNAL POWER WHEN THE
BATTERY PACK IS REMOVED.
ONLY REMOVE THE BATTERY PACK WHEN THE
UNIT IS POWERED OFF AND UNPLUGGED FROM
THE POWER SUPPLY. DO NOT ATTEMPT TOREMOVE THE BATTERY PACK WHILE THE UNIT
IS STILL ON OR PLUGGED INTO THE POWER
SUPPLY.

PN MN232-C4 L | User Manual EN - 11

FreeStyle Comfort
Specifications
FreeStyle Comfort
Dimensions (with Single Battery)

10.0 x 7.3 x 3.1 in (25.4 x 18.5 x 7.9 cm)

Dimensions (with Double Battery)

11.0 x 7.3 x 3.1 in (27.9 x 18.5 x 7.9 cm)

Weight (with Single Battery Pack)

5 lbs (2.3 kg)

Weight (with Double Battery Pack)

6 lbs (2.7 kg)

Nominal Sound Level***

39,93 dB(A) ± 0,42 dB(A) @ 2

Flow Settings

Pulse Setting 1, 2, 3, 4, 5

O2 Output Setting

Setting 1 - 210 mL/min
Setting 2 - 420 mL/min
Setting 3 - 630 mL/min
Setting 4 - 840 mL/min
Setting 5 - 1050 mL/min([@20bpm STPD] All Volumes +/- 15%)

Oxygen Concentration*
Oxygen Monitor
Maximum Outlet Pressure
Dosing Sensitivity

90% (+5.5% / -3%)
Yes
<30 PSI> -0.1 cm H2O (adjustable)

AC Power 100–240 VAC, 50-60Hz, 120W Max
DC Power 11–18 VDC (10 max amp)
Battery Type Lithium Ion
Battery Capacity

Single cell: 6700 mAh; Double cell: 13400 mAh

Battery Re-Charge Time Single Battery: 3.5 hours
Double Battery: 6.0 hours
Battery Pack Durations

Single Battery Duration: Setting 2 - up to 4 hours
Double Battery Duration: Setting 2 - up to 8 hours

Operating Temperature**

41°F to 104°F (5°C to 40°C)

Transportation and Storage Temp. (Unit) -13°F to +158°F (-25°C to +70°C) up to 10000 ft (3048 m)
Transportation and Storage Temp. (Battery) -13°F to +158°F (-25°C to +70°C) up to 10000 ft (3048 m)
Operating Altitude**-1250 to 10,000 ft (-381 to 3048 m) (tested to 700 – 1060 hPa)

Operating Humidity

15 - 95% relative humidity (non-condensing)

Storage Humidity

0 - 90% Non-Condensing

Limited Warranty

See Warranty Statement* Based on an atmospheric pressure of 14.7 psi (101 kPa) at 70°F (21°C)** The unit will function in all environments as long as the environment is with the specified temperature and humidity
ranges. Operating outside of these operational specifications can limit the concentrator’s ability to meet Oxygen Concentration specification at higher liter flow rates. Warm up screen may continue to display until the concentrator has reachedtarget concentration or up to 15 minutes. Continue to use unit during this time.
***Sound level measured per test method Nr. 14-1 10/2018 MDS-Hi.
The expected service life is a minimum of five years.

12 - EN PN MN232-C4 L | User ManualFreeStyle Comfort
Operating Instructions
1. Locate and position the FreeStyle Comfort in aproper location for use.
a. The concentrator should be well-ventilated
so that the air intake filters and air outlet are
not obstructed.
b. The concentrator should be positioned so that
all audible and visual indicators and alarms
can be easily seen and heard.
c. Be sure the air intake filters are in place before operating the unit. If the unit is missing
a filter, contact your equipment provider.
2. Install a Battery Pack in the concentrator.
a. Align the battery with the bottom housingfrom the center of the FreeStyle Comfort.

3. Insert the AC or DC power supply into thepower port receptacle on the front, bottom rightside of the unit.

4. Connect a nasal cannula to the oxygen outlet
barb. Ensure the connection is secure before
operation.
NOTE: Any cannula(s) used other than the recommended may affect performance.

b. Slide the battery inward until you hear aclick to confirm that it has locked in place.

Note: To Equipment Provider: The following oxygen
administration accessories are recommended for use
with the FreeStyle Comfort:• Nasal Cannula with 7 feet (2.1 m) of tubing (Maximum LPM: 6 LPM): CAIRE Part Number 5408-SEQ• Additional recommended accessories information isavailable online at www.caireinc.com.
5. Turn on the FreeStyle Comfort by pressing thepower button and select your prescribed flow
setting utilizing the increase (+) or decrease (-)
flow buttons.

PN MN232-C4 L | User Manual EN - 13

FreeStyle Comfort
Breath
Rate
15
20
25
30
35
40

non-autoSAT Pulse Volume (mL)* Setting
1
2
3
4
5
14.0
10.5
8.4
7.0
6.0
5.3

28.0
21.0
16.8
14.0
12.0
10.5

42.0
31.5
25.2
21.0
18.0
15.8

56.0
42.0
33.6
28.0
24.0
21.0

70.0
52.5
42.0
35.0
30.0
26.3

A brief audible alarm will sound to indicate thedevice has powered on.

8. Turn the FreeStyle Comfort off by pressing thepower button.

Note: When the FreeStyle Comfort is initially turnedon the default flow setting is 2. If the setting ischanged, the unit will power on to the previous flow
setting.

Note: The display screen will enter standby mode anddim after 20 seconds. Press any button to wake thescreen out of standby mode.

6. Position the nasal cannula on your face per the
manufacturer’s instructions provided with thecannula and breathe normally. You should feel apuff of oxygen delivered through the nasal cannula each time you inhale.
7. To change the flow setting, press the increase (+)
or decrease (-) flow buttons to the desired flow
rate. It is normal to hear a difference in sound as
you change the flow settings.

9. For Advanced Features see Menu Navigation
Instructions

SPECIAL PRODUCT FEATURES
autoSAT®
CAIRE’s proprietary autoSAT feature allows the
FreeStyle Comfort to deliver a consistent volumeof oxygen (bolus) per breath, up to a maximum
breath rate. As the user’s breath rate increases
during activities like walking, the autoSAT featureadjusts the device operation to maintain a set bolus
volume.
UltraSense®
This sensitive pulse dose trigger detects thepressure change during your breath, and releases
the pulse of oxygen when you inhale.

Breath
Rate
15
20
25
30
35
40

autoSAT Pulse Volume (mL)*Setting
1
2
3
4
5
10.5
10.5
10.5
10.5
10.5
10.5

21.0
21.0
21.0
21.0
21.0
21.0

31.5
31.5
31.5
31.5
30.0
26.3

*Tolerance is +/- 15%

14 - EN PN MN232-C4 L | User Manual

42.0
42.0
42.0
35.0
30.0
26.3

52.5
52.5
42.0
35.0
30.0
26.3

autoDOSE®
Should a “no breath alarm” occur, this feature willpulse oxygen at 20 breaths per minute at the unit’s
current flow setting until the alarm state is resolved.

FreeStyle Comfort
Power Supplies
The FreeStyle Comfort can be powered in threeways:• Rechargeable Battery Pack
• AC Power• DC Power
Information on the operation of each of these components is in the following sections.
WARNING: USE ONLY THE POWER
SUPPLIES OUTLINED IN THIS USER’SMANUAL. USE OF NON-CAIRE POWER
SUPPLIES AND BATTERY PACKS CANCAUSE DAMAGE TO THE UNIT AND
IMPROPER OPERATION.

REMOVING THE BATTERY

Push the battery release button downward awayfrom the FreeStyle Comfort. While HOLDING thisbutton, slide the battery out by pulling it away fromthe concentrator.
WARNING: THE OXYGEN CONCENTRATOR MUST BE OFF AND DISCONNECTED FROM EXTERNAL POWER BEFOREREMOVING THE BATTERY PACK.

BATTERY PACKThe FreeStyle Comfort can be operated by the rechargeable battery pack which is supplied with thedevice. There are no operational limitations of theconcentrator when running from the battery pack.
All flow rates are available on battery power.

CHECKING BATTERY PACK CHARGEINSTALLING THE BATTERY

Align the battery with the bottom housing from thecenter of the FreeStyle Comfort. Push the battery
inward until you hear a click to confirm that it has
locked in place.

To check the battery charge when it is installed in
the FreeStyle Comfort, reference the battery icon
and the estimated time remaining on the display.
80–100%60–80%

40–60%

20–40%

20%

PN MN232-C4 L | User Manual EN - 15

FreeStyle Comfort
Note: If the power supply is attached, a plug indicator
will show instead of the charge percentage remaining.
To check the battery charge when it is not installedin the FreeStyle Comfort, lay it flat and press the
TEST button. The battery gauge indicator lights(25-100%) will illuminate to the right of the TEST
button to indicate the level of the battery packcharge.

BATTERY OPERATING DURATION

When external AC or DC power is disconnected,
the concentrator will automatically switch over tousing battery power if a charged battery is installed.
The table below shows the typical durations fora new battery pack. Battery pack durations willremain constant regardless of your breath rate.
Pulse
Setting

Single (8-cell)
Battery Duration

Double (16-cell)
Battery Duration1

up to 8 hours

up to 16 hours

2

up to 4 hours

up to 8 hours

3

up to 3 hours

up to 6 hours

4

up to 2.25 hours

up to 4.5 hours

5

up to 2 hours

up to 4 hours

TYPICAL BATTERY RECHARGE TIMEINITIAL CHARGINGThe new battery supplied with your FreeStyle
Comfort is only partially charged when it is shippedfrom the factory. Before using for the first time, you
should fully charge the battery pack using either
AC or DC power.

GENERAL CHARGING INFORMATIONThe FreeStyle Comfort will recharge the battery
pack anytime it is installed in the concentrator andthe concentrator is connected to an external AC or
DC power source (except on an airplane).
Battery is fully charged when it displays 100%.
Alternately, the battery pack may be charged using
the optional desktop charger.

16 - EN PN MN232-C4 L | User Manual

The typical time to fully recharge your battery packfrom a fully discharged battery is 3.5 hours for asingle battery, and 6.0 hours for a double battery.
Note: The battery may not initially start charging if thebattery pack has been fully depleted.

BATTERY SERVICE LIFE

The FreeStyle Comfort batteries are designed tolast 300 charge/discharge cycles.

FreeStyle Comfort
AC & DC Power Supply
The FreeStyle Comfort comes standard with AC
and DC power supplies.
The AC power supply includes the following:a. AC Power brick that operates between
100-240 volts, 50-60Hz.
b. AC Power cord to connect to the power
brick and corresponding AC outlet.

The DC power supply consists of a cord that plugs
directly into the unit and the DC outlet.

AC POWER

2. Connect the AC power cord to the AC power
brick.

AC power is for use at home or wherever standard
AC outlet power is available. The FreeStyle Comfort is fully functional on AC power. The FreeStyle
Comfort will also recharge the battery pack any
time AC power is available (except on an airplane).
The battery pack will recharge whether the FreeStyle Comfort is powered on or off.
WARNING: DO NOT CONNECT EXTERNAL POWER WITHOUT A BATTERYPACK INSTALLED IN THE OXYGENCONCENTRATOR.
To connect the FreeStyle Comfort to AC Power:
1. Connect the AC power brick to the power
connector inlet on the front, right bottom of the
FreeStyle Comfort. Be sure to push inward sothat the connection is secure.

3. Connect the opposite end of the AC power cord
to the wall or other appropriate electrical outlet.

PN MN232-C4 L | User Manual EN - 17

FreeStyle Comfort
4. Turn on the FreeStyle Comfort by pressing thepower button and select your prescribed flow
setting utilizing the increase (+) or decrease (-)
flow buttons.

4. Turn on the FreeStyle Comfort by pressing thepower button and select your prescribed flow
setting utilizing the increase (+) or decrease (-)
flow buttons.
Note: If the battery pack needs to be recharged, thebattery icon on the display will have a single orangebar.

DC POWER

DC power is for use in the accessory outlets ofautomobiles, boat, or other motor vehicles. The
FreeStyle Comfort is fully functional for use on DC
power. The FreeStyle Comfort will also recharge
the battery pack if needed any time DC power isavailable. The battery pack will recharge both if theunit is off, or on and in use.
WARNING: DO NOT CONNECT EXTERNAL POWER WITHOUT A BATTERYPACK INSTALLED IN THE OXYGENCONCENTRATOR.
1. Turn on your motor vehicle.
2. Connect the DC power supply to the power port
receptacle front, bottom right side of the FreeStyle Comfort. Be sure to push inward so that theconnection is secure.

3. Connect the opposite end of the DC power cord
to the automobile or other DC accessory outlet.
18 - EN PN MN232-C4 L | User Manual

CAUTION: To prevent possible fire or
overheating, ensure the 12V DC power
outlet is free of debris and the adapter
plug of the DC Power Supply fits properly and remains securely in place.
CAUTION: DC power supply metal surfaces canbecome VERY HOT when in use due to highcurrent. Allow sufficient time for surfaces to cool
before touching.
CAUTION: Do not use the DC power supply withany extensions, splitters, or adapters. Use of unapproved accessories may result in them overheating
or being damaged.
CAUTION: Attaching the device to a 12V DC power
outlet can deplete the vehicles battery if the engine
is not running, even if the device is turned off.
CAUTION: Do not jump-start a vehicle while the DC
power supply is connected.

FreeStyle Comfort
Cleaning, Care & Routine
Maintenance
ROUTINE MAINTENANCE

Preventative MaintenanceThe device is maintenance free. Filters inside thedevice are intended to last the lifetime of the device. If the device is not working properly, the unit
should be returned to provider for repair. Servicing
of internal components inside of the cabinet of
the FreeStyle Comfort must be performed by theoxygen provider.
The operator of the unit is responsible for thefollowing maintenance procedure.
• Weekly – Clean the Air Intake Filters
WARNING: DO NOT ATTEMPT TO OPEN
OR REMOVE THE PLASTIC COVERS OF
THE OXYGEN CONCENTRATOR. THE
UNIT SHOULD ONLY BE OPENED BY AQUALIFIED SERVICE TECHNICIAN.

AIR INTAKE FILTERS

Air enters the FreeStyle Comfort through air intake
filters located on the front, bottom of the oxygen
concentrator. These filters prevent dust and otherlarge particles in the air from entering the unit.
At least once per week, the air intake filters should
be cleaned. Your oxygen provider may advise you
to clean them more frequently, depending upon theoperating conditions. Follow these steps to properly
clean the air intake filters.
1. Remove the two filters from the front of the
FreeStyle Comfort by pushing up on the corresponding tab and pulling outward.

WARNING: DO NOT OPERATE THE
OXYGEN CONCENTRATOR WITHOUTTHE AIR INTAKE FILTER IN PLACE.

BATTERY PACK CARE

Do not allow the battery pack to be exposed toliquids.
Do not attempt to disassemble or open the battery
pack.
Do not expose the battery to open flames.
Do not drop the battery pack.
Keep the battery pack away from children.
If your battery pack appears damaged, contact youroxygen provide before use.
Do not place the battery in or near a microwave orother cooking appliances.

BATTERY PACK STORAGE

The oxygen concentrator battery does not need to
be fully discharged before recharging. It is recommended to charge the oxygen concentrator battery
after each use.
Do not leave the battery in a location exposed toexcessive heat, such as in direct sunlight or in a car.
Doing so could cause the battery’s performance andlife to deteriorate. It may also cause the battery to
generate heat, smoke, catch fire, or explode. The
battery pack should be stored in a stable environment characterized by low-humidity (less than 70%
RH) and free of corrosive gasses. To prevent rust,
avoid conditions that can create condensation suchas rapid fluctuations in the ambient temperature.
For storage of a month or more, a discharged or
partial charged state of charge between 50% and
75% (two to three LEDs) is recommended. Once
charged, remove the battery from the oxygen
concentrator.

2. Wash both filters in a warm solution of soap andwater.
3. Rinse the filters thoroughly and remove excesswater with a soft, lint-free cloth. Ensure the filters
are completely dry before replacing it.
4. Re-install the clean, dry filters.
PN MN232-C4 L | User Manual EN - 19

FreeStyle Comfort
CLEANING

Turn OFF the unit and disconnect from AC or DC
power before any cleaning. DO NOT spray theouter case directly. Use a damp (not wet) clothor sponge. Spray the cloth or sponge with a mildhousehold cleaner solution to clean the cabinet andpower supplies. Proceed as directed by the cleaner
manufacturer.
WARNING: TURN OFF THE UNIT AND
DISCONNECT THE POWER CORD FROMTHE ELECTRICAL OUTLET BEFOREYOU CLEAN THE UNIT TO PREVENT
ACCIDENTAL ELECTRICAL SHOCK AND
BURN HAZARD. ONLY YOUR EQUIPMENT PROVIDER OR A QUALIFIED SERVICE TECHNICIAN SHOULD REMOVETHE COVERS OR SERVICE THE UNIT.
WARNING: CARE SHOULD BE TAKEN TOPREVENT THE OXYGEN CONCENTRATOR AND
THE BATTERY PACK FROM GETTING WET OR ALLOWING FLUIDS TO ENTER THE UNIT. THIS CANCAUSE A MALFUNCTION OR SHUT DOWN, AND
CAUSE AN INCREASED RISK FOR ELECTRICALSHOCK OR BURNS.
WARNING: DO NOT USE LIQUID DIRECTLY ON
THE UNIT. A LIST OF UNDESIRABLE CHEMICALAGENTS INCLUDES BUT IS NOT LIMITED TO THE
FOLLOWING: ALCOHOL AND ALCOHOL-BASEDPRODUCTS, CONCENTRATED CHLORINE-BASEDPRODUCTS (ETHYLENE CHLORIDE), AND OILBASED PRODUCTS. THESE ARE NOT TO BE USEDTO CLEAN THE PLASTIC HOUSING ON THE CONCENTRATOR, AS THEY CAN DAMAGE THE UNIT.
WARNING: CLEAN THE CABINET, CONTROLPANEL, AND POWER CORD ONLY WITH A MILD
HOUSEHOLD CLEANER APPLIED WITH A DAMP(NOT WET) CLOTH OR SPONGE, AND THEN
WIPE ALL SURFACES DRY. DO NOT ALLOW ANY
LIQUID TO GET INSIDE THE CONCENTRATOR. PAY
SPECIAL ATTENTION TO THE OXYGEN OUTLETFOR THE CANNULA CONNECTION TO MAKE
SURE IT REMAINS FREE OF DUST, WATER, AND
PARTICLES.

20 - EN PN MN232-C4 L | User Manual

WARNING: USE A DAMP (NOT WET) CLOTH ORSPONGE TO CLEAN THE BATTERY PACK. FIRSTSPRAY THE CLOTH OR SPONGE WITH A MILD
HOUSEHOLD CLEANER AND CLEAN THE BATTERY PACK HOUSING.
WARNING: DO NOT CLEAN WITH AUTOMATED
WASHING SYSTEM.

CANNULA REPLACEMENT

Always follow the cannula manufacturer’s instructions for proper use. Replace the nasal cannula or
oxygen tubing every 3 to 6 months (if daily use, 3months) as per manufacturer’s recommendation
as recommended by the cannula manufacturer oryour oxygen provider. Your physician or oxygen provider will provide you with cleaning andreplacement instructions.
Additional supplies for replacement are available
through your oxygen provider.

FreeStyle Comfort
Alarm Conditions
Priority #Alarm
Description

1

System
Malfunction
Battery High
Temperature

Display

AudibleSignalMore Details / Suggested Action

1 beepevery 20
sec

Unit will alarm when battery temperature isapproaching safety shutdown threshold.
If shutdown threshold is reached, unit will
shut down. If external power is connected,
alarm screen will display and audible alarm
will continue for 2 min.
Test with battery at room temperature. If
problem persists, contact provider.

2

System
MalfunctionHigh
Temperature

1 beepevery 20
sec

Unit will alarm when internal temperature isapproaching shutdown threshold.
If shutdown threshold is reached, unit will
shut down and alarm screen will display
and audible alarm will continue for 2 min.
Ensure there are no blockages on any
vents of the unit. If problem persists,
contact provider.

3

System
Malfunction
High
Pressure(High System
Pressure orCompressor
Fault)

1 beepevery 20
sec

4

System
Malfunction
Flow Rate(OCD Valve
Failure,Obstructionof GasPathway)

1 beepevery 30
sec

If problem persists, contact provider.

5

Low Battery
Shut Down

1 beepevery 30
sec

Alarms until battery runs out of power andunit shuts down.

If shutdown threshold is reached, unit will
shut down and alarm screen will display
and audible alarm will continue for 2 min.
If problem persists, contact provider.

Charge battery.

PN MN232-C4 L | User Manual EN - 21

FreeStyle Comfort
Priority #Alarm
Description

6

No Breath
Detected

Display

AudibleSignalMore Details

1 beepevery 10min

No breath detected for 15 seconds.
Activated after first breath or warm up
screen is off, whichever comes sooner.
Check cannula for kinks or leaks. Increase
the sensitivity setting. If problem persists,
contact provider.

7

Warm Up

none

Unit will start by auto-dosing until a breathis detected or Warm up ends, whicheveris sooner.

8

Low
Concentration

1 beepevery 10
min, forup to 30min

If problem persists, contact provider.

9

Low Battery
Warning

1 beep,no repeat

Charge battery.

10

Breathing
Rate
Exceeded

1 beep,no repeat

Breathing Rate exceeds 40 bpm.

11

System
Malfunction
Fan Failurenone

If problem persists, contact provider.

12

System
Malfunction
Low Pressurenone

If problem persists, contact provider.

13

System
Malfunction
Low Temp

none

Allow unit to warm up to room temperature.
If problem persists, contact provider.

22 - EN PN MN232-C4 L | User Manual

If dose is being falsely triggered, decrease
the sensitivity setting. If problem persists,
contact provider.

FreeStyle Comfort
Priority #Alarm
Description

Info

Start up

Display

AudibleSignalMore Details

1 infobeep

Screen displays at power-on and you
should hear one beep to confirm buzzer isfunctioning
Start up screen display current firmware
revision in bottom right corner.

Info

Normal Run
Screen in
autoSAT
Mode (noDC Powerconnected)

none

Info

Normal Run
Screen with
autoSATdisabled (NoDC Powerconnected)

None

Info

SettingChanged

1 infobeep

InfoPlug inexternal
Power in theoff condition

none

Info

Plug inexternal
power in the
run condition

none

Info

Unplugexternal
power in theoff Condition

1 infobeep

Display CAIRE screen for 2 seconds,
followed by Battery & Plug status for 5 sec
before screen turns off

Display CAIRE screen for 2 seconds andthen turn display off.

PN MN232-C4 L | User Manual EN - 23

FreeStyle Comfort
Priority #Alarm
Description

DisplayInfo

Unplugexternal
power in the
run Condition(with battery
present)

1 infobeep

Info

In Standby
Mode - whenup or down
button ispresseddisplay showsbattery andcharging
state

none

Info

Airplane
Mode on battery power

1 longbeep

Info

Airplane
Mode while
plugged in

1 longbeep

24 - EN PN MN232-C4 L | User Manual

AudibleSignalMore Details

Screen will display for 5 sec then turn off

FreeStyle Comfort
Menu Navigation Instructions
You can enter the Menu in Standby or Run mode.
In Run Mode, if an alarm condition occurs themenu will automatically be exited. If no buttons arepressed for 20 seconds, the Menu will automatically be exited. In Standby, you may need to press
the Menu button twice to wake unit, then enter the
Menu.
There are four Menu screens:
1. Status screen: Displays Firmware rev number,run time, battery discharge cycles

Airplane Mode

2. Airplane mode (ON/OFF)
3. Sensitivity: adjustable (1-5)
4. autoSAT (ON/OFF)

Sensitivty

Example Status Screen

SETTINGS MENU OPERATION
Settings menu is entered by pressing Menu/SelectbuttonautoSATin Run or Standby modes.

To scroll through the menus, push the.

Status Page

PN MN232-C4 L | User Manual EN - 25

FreeStyle Comfort
STATUS PAGE
The Status Page of the Freestyle Comfort shows
Firmware Revision, Run Hours, and number ofbattery cycles on the attached battery.

To cancel Airplane Mode on, press
CANCEL icon is visible.

while theAirplane Mode On Confirmed

Status Page

Run hours are shown in whole hours.
Battery Discharges is the number of cycles theattached battery has been drained and recharged.

AIRPLANE MODE
The FreeStyle Comfort ships with Airplane Modeturned off.

To turn Airplane Mode off, repeat above while
toggling to OFF.
Pressing
when the screen show the “Next”
symbol will move the menu page to the next screen
option.
Airplane Mode is intended for use in Run Mode.
Airplane Mode will be automatically disabled if theunit is left in Standby on battery power.
Note: See section Activate Airplane Mode for more
information about Airplane Mode.

SENSITIVITY MODE

Airplane Mode Main Screen

To turn the mode on, press
while the CHANGE
icon is visible. The value will toggle to the blue
word “ON”.

The FreeStyle Comfort ships with a default Sensitivity Mode setting of 2 from the factory and should
not require adjustment under normal conditions.
Sensitivity Setting can be adjusted between values
1-5, where 1 is the least sensitive, 5 is the mostsensitive.

Sensitivity Mode Screen at Setting 2
Airplane Mode On Selected

To confirm Airplane Mode on, press
while theSAVE icon is visible. The value will toggle to theblack word “ON”.

26 - EN PN MN232-C4 L | User ManualFreeStyle Comfort
To adjust the Sensitivity Mode, press
while theCHANGE icon is visible. The value will toggle tothe next blue number.

Sensitivity Mode Setting 3 Selected

To confirm Setting 3 in Sensitivity Mode, press
while the SAVE icon is visible. The value will
toggle to the black “3”.

AUTOSAT MODE
The FreeStyle Comfort ships with autoSAT Modeturned on.

autoSAT Mode Main Screen

To turn the mode off, press
while theCHANGE icon is visible. The value will toggle tothe blue word “OFF”.

To cancel the Sensitivity Mode change, press
while the CANCEL icon is visible.

autoSAT Mode Off Selected

Sensitivity Mode Setting 3 Confirmed

To confirm autoSAT Mode off, press
while theSAVE icon is visible. The value will toggle to theblack word “OFF”.
To cancel autoSAT Mode on, press
CANCEL icon is visible.

while theSensitivity Mode Setting 5 (Max) Confirmed

Pressing
when the screen show the “Next”
symbol will move the menu page to the next screen
option.

autoSAT Mode Off Confirmed

To turn autoSAT Mode on, repeat above while
toggling to ON.
Pressing
when the screen show the “Next”
symbol will move the menu page to the next screen
option.

PN MN232-C4 L | User Manual EN - 27

FreeStyle Comfort
Troubleshooting
Problem

Probable Cause

Solution

FreeStyle Comfort displays
alarm and produces
intermittent beep.

Refer to Alarm Conditions
table.

Refer to Alarm Conditions table.

Oxygen concentrator does
not turn on.

Battery is discharged.

Power the unit through the AC or DC outlet.

Not connected to external
power.

Ensure that external connects are secure.

General malfunction.

Perform a Full Power Cycle by removing thebattery and disconnecting any AC or DC external
power source. Then reinstall battery, and thepower cycle is complete. Reconnect external
power source if desired.
Contact your Equipment Provider, and change toanother source of oxygen as necessary.

All other problems.

Perform a Full Power Cycle by removing thebattery and disconnecting any AC or DC external
power source. Then reinstall battery, and thepower cycle is complete. Reconnect external
power source if desired.
Change to another source of oxygen as
available, and contact your Equipment Provider.

Note: Some alarms will cause the unit to shut down.
Note: During a total power loss, the alarm system resets (clears any previous alarms) and will only display an
alarm if the condition still exists once power is restored.

28 - EN PN MN232-C4 L | User ManualFreeStyle Comfort
Travel Information

1. Turn POC On using Battery Power ONLY, whenno external power source is connected.

The FreeStyle Comfort meets FAA guidelines foruse on-board commercial aircraft. Prior to any
travel (airline, train, cruise, etc.) you should:• Ensure the concentrator is in good operation
condition.
• Contact your carrier at least 48 hours prior to
departure to inform them you will be travelingwith a FreeStyle Comfort.
It is suggested to always carry extra batteries withyou on trips. For airline travel, please allow enoughbattery life to account for ground time (preflightcheck in, security, etc.) and layovers. Most airlines
require that you have an adequate number of fully
charged battery packs to power the device for atleast 150% of the expected maximum trip duration(flight times, ground time before and after theflight, connections, and unexpected delays).

2. Enable Airplane Mode using one of the following methods:
a. Follow directions to enable Airplane Mode in
Menu Navigation Instructions section.

Note: If flying, it is recommended that extra batteries
are carried on in a secure carrying bag.

FAA REGULATION

Spare batteries must be individually protected toprevent short circuits by placement in the originalretail packaging or by otherwise insulating theterminals, e.g. by taping over exposed terminals
or placing each battery in a separate plastic bag orprotective pouch, and carried in carry-on baggage
only.

b. Press and hold both Up and Down buttons for
5 seconds until you hear a long beep and see theAirplane symbol appear.

ACTIVATE AIRPLANE MODE
Battery Power
POC can be used on an aircraft per normal operating procedures using Battery Power ONLY, whenno external power source is connected
External PowerOn an airplane, if an electrical outlet is available,
operable, and its location is appropriate for use
as a POC power source, it may serve as a backup to POC batteries. In this case, the following
recommended procedure regarding the transition
from battery to aircraft electrical power must be
followed:

3. When the airplane symbol is displayed, plug in
the POC.

Note: Due to aircraft limitations, the batteries do not
charge in Airplane Mode.

PN MN232-C4 L | User Manual EN - 29

FreeStyle Comfort
Accessories

INSERTING UNIT INTO CARRYING BAG

For proper performance and safety, use only theselisted accessories supplied by CAIRE throughyour oxygen provider. Use of accessories not listed
below could adversely affect the performance and/or safety of the concentrator.

1. Insert unit into bottom of bag, properly aligningfilters and battery openings.

FreeStyle Comfort Standard Accessories
Carrying Bag

MI396-1

Backpack

MI459-1

Shoulder Strap

MI397-1

AC Power Supply (All Cords
Included)

PW036-1SDC Power Cord

CD041-1

External Gross Particle Filter (Right)

FI226-1

External Gross Particle Filter (Left)

FI227-1

2. Secure zipper around bottom of pouch.

Optional Accessories Available
Additional 8-Cell Battery Pack

BT034-1S

16-Cell Battery Pack

BT035-1SDesktop Battery Pack Charger

BT036-1

Carry-all Accessory Bag

MI372-2

Note: Additional options may be available for country-specific power cords where noted above. Contact
CAIRE or your oxygen provider if alternate options areneeded for order.

SHOULDER STRAP INSTALLATION
1. Shoulder strap can be attached to either the unit
directly or the carrying bag.
2. Clip strap onto the ring on the unit or bag.

Carrying Bag Instructions
To insert and secure your FreeStyle Comfort in the
carrying bag, utilize the following instructions.

30 - EN PN MN232-C4 L | User ManualFreeStyle Comfort
Disposal and Reuse
Always return FreeStyle Comfort, including allcomponents, to your equipment provider for proper
disposal. You can also contact your local city or
town offices for instructions on proper disposal of
the battery. The unit can be reused between people
but must be returned to equipment provider for
reprocessing.

WEEE and RoHS
This symbol is to remind the equipment
owners to return it to a recycling facility at theend of its life, per Waste Electrical and Electronic
Equipment (WEEE) Directive.
Our products will comply with the restriction ofHazardous Substances (RoHS) directive. They will
not contain more than trace amounts of lead orother hazardous material content.

Materials in Direct Contact with the
Operator• Concentrator Casing: PC/ABS• Concentrator Control Panel: Polyester EBG7 orequivalent• Oxygen Outlet, Nozzle: Aluminum• Unit Label: Lexan• Cord Connectors: Polycarbonate/Vinyl chloride• Power Cord: PVC, Metal• Power Supply: Lexan 940(Polycarbonate)• Battery Pack: PC/ABS• Battery Pack, Power Supply Labels: Polyester
film• Concentrator Carrying Case: Cordura, Neoprene,
Nylon Lining• Strap: Polypropylene, Acetyl, Vinyl, Cordura,
Neoprene

CAUTION: For proper disposal,
contact your equipment provider or
local government agency for disposal
instructions
CAUTION: Always dispose of molecular sieve inaccordance with all federal, state, and local environmental control regulations.

PN MN232-C4 L | User Manual EN - 31

FreeStyle Comfort
EMC Testing
Guidance and Manufacturer’s Declaration­—Electromagnetic Emissions
The FreeStyle Comfort is intended for use in the electromagnetic environment specified below. The customer or theuser of the FreeStyle Comfort should assure that it is used in such an environment.
Emissions Test
RF emissions
CISPR 11
RF emissions
CISPR 11
Harmonic emissions
IEC 61000-3-2
Voltage fluctuations/
flicker emissions
IEC 61000-3-3ComplianceElectromagnetic Environment - GuidanceGroup 1The FreeStyle Comfort uses RF energy only for its internal function. Therefore, its RF emissions are very low and are not likely tocause any interference in nearby electronic equipment.

Class BClass A
Complies

32 - EN PN MN232-C4 L | User Manual

The FreeStyle Comfort is suitable for use in all establishments,
including domestic establishments and those directly connected tothe public low-voltage power supply network that supplies buildings used for domestic purposes.

FreeStyle Comfort
Guidance and Manufacturer’s Declaration—Electromagnetic Immunity

The FreeStyle Comfort is intended for use in the electromagnetic environment specified below. The customer or theuser of the FreeStyle Comfort should assure that it is used in such an environment.
Immunity test
IEC 60601 test level
Compliance level
Electromagnetic environment –guidance
Electromagnetic
Floors should be wood, concrete orenvironment – guid- ±8 kV contact±8 kV contact
ceramic tile. If floors are covered withance
synthetic material, the relative±15 kV air discharge±15 kV air discharge
humidity should be at least 30 %.
IEC 61000-4-2
Electrical fasttransient/burst
IEC 61000-4-4

Surge
IEC 61000-4-5±2 kV for power supplylines±1 kV for input/output
lines±1 kV line(s) to line(s)±2 kV for power supply
lines
N/A

±2kV common mode on ±1 kV line(s) to line(s)AC lines±1kV differential on
AC lines

Mains power quality should be that of atypical professional healthcare
facility and home healthcare
environments.

Mains power quality should be that of atypical professional healthcare
facility and home healthcare
environments.

±2kV common mode on
outdoor I/O lines
0% UT for 0.5 cycle (0°, 0% UT for 0.5 cycle (0°,45°, 90°, 135°, 180°,
45°, 90°, 135°, 180°,
Mains power quality should be that of
225°, 270°, 315°)
225°, 270°, 315°)
a typical professional healthcare facility
Voltage dips, shortand home healthcare environments. If
00interruptions and
0% UT for 1 cycle (0 )0% UT for 1 cycle (0 ) the user of the FreeStyle Comfort revoltage variations
quires continued operation during poweron power supply
mains interruptions, it is recommended
70%
U
70%
U
(30%
dip
in
U
)
(30%
dip
in
U
)
input lines IEC
T
T
T
T
that the FreeStyle
0
0
)
)
for
25/30
cycles(0for
25/30
cycles(061000-4-11
Comfort is powered from anuninterruptible power supply or a battery.
0% UT for 250/300
0% UT for 250/300
00cycles (0 )cycles (0 )Power frequency magnetic fields should
Power frequency
be at levels characteristic of a typical(50/60 Hz) magnetic 3 A / m
3A/mlocation in a professional healthcare
field IEC 61000-4-8
facility and home healthcare environments.
NOTE UT is the A.C. mains voltage prior to application of the test level.
NOTE In some cases, the Freestyle Comfort may reset after a significant Electrostatic Discharge, and may requirea Full Power Cycle to resume normal operation. See Troubleshooting section for more details.

PN MN232-C4 L | User Manual EN - 33

FreeStyle Comfort
Guidance and Manufacturer’s Declaration–Electromagnetic Immunity

The FreeStyle Comfort is intended for use in the electromagnetic environment specified below. The customer or theuser of the FreeStyle Comfort should assure that it is used in such an environment.
Immunity test
IEC 60601 test level
Compliance level
Electromagnetic environment –guidance
10 Vrms
Portable and mobile RF communications
10 Vrms
Conducted RF
equipment should be used no closer
150 kHz to 80 MHz
IEC 61000-4-6
to any part of the FreeStyle Comfort,
including cables, than the recommended
separation distance calculated from theequation applicable to the frequency of
10 V/m
10 V/mRadiated RF
the transmitter.
80 MHz to 2700 MHz
IEC 61000-4-3
Recommended separation distanced  1.2 P
d  1.2 P = 80 MHz to 800 MHz
d  1.2 P = 800 MHz to 2700 MHz
where P is the maximum output power
rating of the transmitter in watts (W)according to the transmitter manufacturer
and d is the recommended separation
distance in meters (m).
Field strengths from fixed RF transmitters, as determined by an electromagnetic site survey,a should be less thanthe compliance level in each frequency
range. b
Interference may occur in the vicinityof equipment marked with the following
symbol:NOTE 1 At 80 MHz and 800 MHz, the higher frequency range applies.
NOTE 2 These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorption andreflection from structures, objects and people.
aField strengths from fixed transmitters, such as base stations for radio (cellular/cordless) telephones and land
mobile radios, amateur radio, AM and FM radio broadcast and TV broadcast cannot be predicted theoretically withaccuracy. To assess the electromagnetic environment due to fixed RF transmitters, an electromagnetic site survey
should be considered. If the measured field strength in the location in which the FreeStyle Comfort is used exceeds
the applicable RF compliance level above, the FreeStyle Comfort should be observed to verify normal operation. If
abnormal performance is observed, additional measures may be necessary, such as re-orienting or relocating the
FreeStyle Comfort.
bOver the frequency range 150 kHz to 80 MHz, field strengths should be less than 10 V/m.

34 - EN PN MN232-C4 L | User ManualFreeStyle Comfort
Recommended separation distances between portable and mobile RFcommunications equipment and the FreeStyle Comfort

The FreeStyle Comfort is intended for use in an electromagnetic environment in which radiated RF disturbances
are controlled. The customer or the user of the FreeStyle Comfort can help prevent electromagnetic interference by
maintaining a minimum distance between portable and mobile RF communications equipment (transmitters) andthe FreeStyle Comfort as recommended below, according to the maximum output power of the communications
equipment.
Rated maximum
Separation distance according to frequency of transmitter
output power ofmtransmitter
150 kHz to 80 MHz
80 MHz to 800 MHz
800 MHz to 2,5 GHzd  1.2 P
d  1.2 P
d  1.2 P
W
0,01
0.12
0.12
0.23
0,1
0.38
0.38
0.73
1
1.2
1.2
2.3
10
3.8
3.8
7.3
100
12
12
23
For transmitters rated at a maximum output power not listed above, the recommended separation distance d inmeters (m) can be estimated using the equation applicable to the frequency of the transmitter, where P is the maximum output power rating of the transmitter in watts (W) according to the transmitter manufacturer.
NOTE 1 At 80 MHz and 800 MHz, the separation distance for the higher frequency range applies.
NOTE 2 These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorption andreflection from structures, objects and people.

Classification

Type of protection against electric shock:Class II
Protection from electric shock is			achieved by double insulation.
Degree of protection against electric shock:Type BF Equipment providing a particular		 degree of protection against electric
		 shock regarding		 1) allowable leakage current;		 2) reliability of protective earth connection
		 (if present).
		 Not intended for direct cardiac application.
Independent testing for Medical Electrical Equipment Standard:
Tested to be in compliance with
• IEC 60601-1 edition 3.1: Medical Electricalequipment - Part 1 General Requirements forsafety.
• CAN/CSA C22.2 No. 60601-1-8 edition
2.1, Medical Electrical Equipment – Part 1:
General Requirements for Safety
Protection against potential electromagnetic orother interference between the equipment and otherdevices.
• Tested to be in compliance with EN 60601-1-2/IEC 60601-1-2: 4th Edition.

• Tested to be in compliance with RTCA/DO160
Section 21 Category M.
• CISPR 11 / EN 55011 Class B Group 1,“Industrial, Scientific, and Medical (ISM)
Equipment”• FCC Part 15, Subpart B – Class B
Unintentional Radiators
IP22 - Protection against ingress of fingers orsimilar objects greater than 12.5 mm diameter.
Vertically dripping water shall have no harmful
effect when the enclosure is tilted at an angle of
15° from its normal position.
Method of cleaning and infection control allowed:
Please refer to “Cleaning, Care, and Proper
Maintenance” section of this FreeStyle User
Manual.
Degree of safety of application in the presence offlammable anesthetic gases:
Equipment not suited for such application.
Mode of operation: Continuous duty.

PN MN232-C4 L | User Manual EN - 35

FreeStyle Comfort
Commandes d’utilisation et indicateurs d’état du système
ISO 7000
Manuel d’utilisation ; instructions
d’utilisation. Nº enreg. 1641

ISO 7010 : symboles graphiques – couleursde sécurité et étiquettes de sécurité – étiquettesde sécurité enregistrées

Le pays et la date de fabrication.
Le « CC » identifie le code pays à deux
lettres du pays de fabrication. La date defabrication est au format AAAA-MM-JJ.
Nº enreg. 6049

Le manuel d’instruction doit être lu.
Nº enreg. M002
Tenez l’unité à l’écart des flammes nues,
du feu, des étincelles. Source d’incendie
ouverte interdite et interdiction defumer. Nº enreg. P003
Ne fumez pas à proximité de l’appareilou pendant son fonctionnement.
Nº enreg. P002

Maintenez l’unité à l’abri de l’humidité.
Nº enreg. 0626
Limite d'empilement par nombre,où « n » indique le nombre maximum
d'unités autorisées. Nº enreg. 2403
Référence catalogue. Nº enreg. 2493

Avertissement. Nº enreg. W001
CEI 60417

Numéro de série. Nº enreg. 2498

Pièce appliquée de type BF (degré
de protection contre les déchargesélectriques). Nº enreg. 5333

Plage de température de fonctionnement.
Nº enreg. 0632

Équipement de classe II à double
isolation N° enreg. 5172

Plage d'humidité de stockage.
Nº enreg. 2620
Limite de température atmosphérique.
Nº enreg. 2621
Nom et adresse du fabricant.
Nº enreg. 3082

Règlement UE 2017/745
Représentant agréé au sein de l’Union
européenne
Si l’étiquette d’identification uniquedes dispositifs (IUD) porte le symbole
CE####, le dispositif est conforme
aux exigences relatives aux dispositifs
médicaux. Le symbole CE#### indiquele numéro de l’organisme notifié.
Symboles supplémentaires

Haut. Nº enreg. 0623

Tenez l’unité éloignée des matériaux,des huiles et des graisses inflammables.

Fragile, à manipuler avec soin.
Nº enreg. 0621

Ne pas démonter.

Contient des substances dangereuses.
Nº enreg. 3723

Augmenter le débit.

Importateur. Nº enreg. 3725

Diminuer le débit.

Sortie de l’O2

Bouton Menu.
Interrupteur. Nº enreg. 5009
Conforme
à lacatégorie M
de lasection 21
de la norme
RTCA
DO-160.

36 - FR PN MN232-C4 L | Manuel d’utilisation

Le fabricant de ce POC a déterminé
que ce dispositif est conforme à tous
les critères d'acceptation applicables
de la FAA et à l'utilisation à bord desaéronefs.

FreeStyle Comfort
Le fabricant de ce POC a déterminé
que ce dispositif est conforme à tous lescritères d'acceptation applicables de la
FAA et à l'utilisation à bord des aéronefs.

Directive du Conseil européen 2012/19/UE
DEEE

Ce symbole a pour objet d’inciterle propriétaire de l’équipement à leretourner à un centre de recyclage à lafin de son cycle de vie, conformément
à la directive relative aux déchetsd’équipements électriques etélectroniques (DEEE). Nos produitssont conformes à la Directive sur lalimitation des substances dangereuses(RoHS). Ils ne contiendront pas plus
qu’une quantité infime de plomb ou
d’autres substances dangereuses.

Symbole de recyclage
Représentant agréé en Suisse (ADD)
Si le dispositif porte la marque
UKCA indiquée avec UKCA####indiquant le numéro de l’organisme
notifié, ce dispositif est conforme
à la réglementation UKCA.
21 CFR 801.15
La législation fédérale restreint la ventede cet appareil par un médecin ou sur
ordonnance de celui-ci.
Certification ETL

CEI 60601-1-8
Alarme technique priorité faible
ISO 15223-1

Certifié pour les États-Unis et le Canada,conformément aux normes desÉtats-Unis et du Canada en vigueur
Contient l'ID FCC : WAP2001
Avertissement de la FCC
Ce dispositif peut contenir un module
CYBLE-022001-00, y compris
l’antenne 2450AT18B100 de Johnson
Technology, conforme avec l’article 15 de laréglementation FCC. Le dispositif est conforme aux
exigences relatives à l’approbation de l’émetteur
modulaire tel que décrit dans l’avertissement
public de la FCC DA00-1407. Le fonctionnement
de l’émetteur est soumis aux deux conditions
suivantes : (1) ce dispositif ne peut provoquerd’interférences nuisibles, et (2) ce dispositif doit
accepter toute interférence reçue, notamment
les interférences susceptibles d’entraîner unfonctionnement non souhaité. Ce produit peut êtreprotégé par un ou plusieurs brevets américains ou
internationaux. Consultez notre site Web ci-dessous
pour voir la liste des brevets. Brevet :www.caireinc.com/corporate/patents
Symboles graphiques à utiliser sur l’équipement
CEI 60417

Dispositif médical. Nº enreg. 5.7.7
N° d'identifiant unique du dispositif
5.7.10A S/NZS 4417.1, AS/NZS 4417.2
Marquage de conformité réglementaire
ACMA – indique la conformité auxréglementations ACMA (ADD)

Technologie sans fil
Types de connexion

SSP, iAP2, GATT

Fréquence

2 402 à 2 480 MHz

Puissance de sortie RF
maximum+4 dBmPortée de fonctionnement

10 m (Classe 2)

Il est recommandé que l’appareil soit placé à une
distance minimum de 0,43 po (1,1 cm) du corps durantle fonctionnement. Ne s’applique pas aux masques, aux
tubes d’air, ou aux accessoires.

Courant continu Nº enreg. 5031
Courant alternatif Nº enreg. 5032
CEI 60601-1
Protection contre les gouttes d’eau

PN MN232-C4 L | Manuel d’utilisation FR - 37

FreeStyle Comfort
Déballage de votre FreeStyle Comfort
Vérifiez que tous lescomposants listés sont inclus
dans l’emballage. En l’absenced’un composant, contactez
immédiatement votre fournisseurd’oxygène.

Accessoires standard• Concentrateur d’oxygène portable• Sac de transport• Bandoulière• Bloc-batterie avec 8 cellules• Alimentation CA• Cordon d’alimentation CA• Alimentation CC• Manuel d’utilisation

Accessoires en option• Pack supplémentaire de batterie
8 cellules• Pack de batterie à 16 cellules• Chargeur de bureau• Sac de transport pour lesaccessoires

Présentation de votre FreeStyle Comfort
Familiarisez-vous avec les principales caractéristiques du FreeStyle Comfort et son interface utilisateur.

Vue avantFixation de la bandoulière

Interfaceutilisateur

Sortied’oxygène
Sortie d’air

Prise d’air/Filtres
Prise d’air/Filtres

Languette pourprise de portd’alimentation
Batterie
Bouton de déblocage
de la batterie
38 - FR PN MN232-C4 L | Manuel d’utilisation

FreeStyle Comfort
Vue de dessus
Écran d’affichage
Augmentationdu débitBouton MARCHE/ARRÊT

Bouton Menu

Diminutiondu débit

Écran d’affichage sur batterie
Réglage du débit

Indicateurde batterie

Indicateur autoSAT
Pourcentage dela durée de vie
restante de labatterie

Écran d’alarme

Indicateurd’administration
du bolus

Écran d’affichage sur alimentation CA/CC

Réglage du débit

Zone du voyant
d’alarme

Indicateurde batterie

Indicateur d’étatde charge

PN MN232-C4 L | Manuel d’utilisation FR - 39

FreeStyle Comfort
Qu’est-ce que le concentrateurd’oxygène ?
L’air que nous respirons contient environ 21 %d’oxygène, 78 % d’azote et 1 % d’autres gaz. Dansle FreeStyle® Comfort®, l’air ambiant de la pièce est
aspiré dans la machine par les prises d’air. Il passe
ensuite au travers d’un matériau absorbant appelétamis moléculaire. Ce matériau sépare l’oxygène del’azote et capture ce dernier pour laisser uniquement
passer l’oxygène. Il en résulte un débit d’une forte
concentration d’oxygène transmis à l’utilisateur.
Le FreeStyle Comfort combine des technologies
avancées de concentration d'oxygène et deconservation de l'oxygène pour créer un appareil
portable qui permet une portabilité et une déambulation
maximales. Grâce à un débit pulsé optimal, l’appareil
détecte rapidement les respirations de l’utilisateur afin
de lui fournir de l’oxygène dès le début de chaqueinhalation. Ce type de débit se révèle parfaitement
adapté pour les activités et les séjours à l’extérieur.
Contrairement à un débit continu, il conservel’oxygène et augmente la durée de vie de la batterie.
Le FreeStyle Comfort est suffisamment léger pourêtre porté en bandoulière. Il fonctionne sur courant
CA (mur), CC (véhicule) ou sur batterie.

Utilisation prévue
Le concentrateur d’oxygène CAIRE FreeStyle
Comfort est indiqué pour l’administration d’oxygène
supplémentaire. Cet appareil n’est pas conçu pour lemaintien des fonctions vitales et ne permet pas nonplus de surveiller les patients.

Pourquoi votre médecin vousprescrit-il de l’oxygène ?
Un grand nombre d’individus souffrent de différentes
maladies cardiaques, pulmonaires ou encore
respiratoires. La majorité d’entre eux peuvent
bénéficier d’une oxygénothérapie complémentaireà domicile, lorsqu’ils voyagent, ou même lorsqu’ils
participent à des activités à l’extérieur.
L'oxygène est un gaz qui constitue 21 % de l'air
ambiant que nous respirons et notre corps a besoind'un approvisionnement régulier pour fonctionnercorrectement. Votre médecin vous a recommandé
un débit ou un réglage à définir afin de traiter votre
affection respiratoire.

40 - FR PN MN232-C4 L | Manuel d’utilisation

Bien que l’oxygène ne constitue en aucun cas
une drogue provoquant un état de dépendance,
une oxygénothérapie non autorisée peut s’avérer
dangereuse. N’oubliez pas de demander conseil à votre
médecin avant d’utiliser ce concentrateur d’oxygène.
Le fournisseur de cet appareil vous montrera comment
régler le débit prescrit.

Profil de l’utilisateur
Les concentrateurs visent à fournir un apportd’oxygène supplémentaire aux utilisateurs pris d’unmalaise découlant d’une maladie qui réduit la capacitédes poumons à distribuer l’oxygène contenu dansl’air dans la circulation sanguine. Les concentrateursd’oxygène portables (COP) ne contiennent pas
d’oxygène. Ils n'ont pas besoin d'un nouveau
remplissage et peuvent être rechargés partout où unealimentation CA ou CC est disponible (sauf dans unavion). Ainsi, l’utilisateur peut facilement l’utiliser à
son domicile et le transporter (à l’intérieur comme àl’extérieur de son domicile), pour un confort de vie
amélioré. L’usage d’un concentrateur requiert uneprescription médicale.
Les patients sont généralement âgés de plus de 65 ans
et souffrent de différentes pathologies respiratoires,notamment la maladie pulmonaire obstructive
chronique (MPOC). En règle générale, les utilisateurssont dotés de capacités intellectuelles suffisantes
pour exprimer leur malaise. Dans le cas contraire,ou lorsque l’utilisateur est incapable de lire et decomprendre les étiquettes apposées sur le concentrateurou encore le mode d’emploi, le concentrateur peut
uniquement être utilisé sous la supervision d’une
personne compétente. Si une gêne est ressentie pendantl’utilisation du concentrateur, il est conseillé aux
utilisateurs de contacter leur prestataire de soins. Il estégalement recommandé aux utilisateurs d’être équipésd’un apport d’oxygène de secours, tel qu’une bouteilled’oxygène, dans le cas d’une coupure électrique
ou d’une panne du concentrateur. L’utilisation
du concentrateur n’exige aucune compétence ni
connaissance particulière.

FreeStyle Comfort
Important !
Les instructions de sécurité se définissent comme
indiqué ci-dessous :AVERTISSEMENT : INFORMATIONSIMPORTANTES DE SÉCURITÉ RELATIVESAUX DANGERS POUVANT CAUSER DESBLESSURES GRAVES.
MISE EN GARDE : informations
importantes pour éviter d'endommager
le FreeStyle Comfort.
Remarque : informations exigeant une attention particulière.

Instructions d’utilisation
AVERTISSEMENT : IL EST TRÈSIMPORTANT DE SÉLECTIONNER LE DÉBITD’OXYGÈNE QUI VOUS A ÉTÉ PRESCRIT.
NE MODIFIEZ PAS LE RÉGLAGE DE DÉBITSANS AVIS MÉDICAL.
AVERTISSEMENT : CE DISPOSITIF N’A PAS ÉTÉTESTÉ NI APPROUVÉ POUR UNE UTILISATION CHEZ
DES PATIENTS PÉDIATRIQUES.
AVERTISSEMENT : CONFORMÉMENT À LA LOIFÉDÉRALE DES ÉTATS-UNIS, LA VENTE ET LA
LOCATION DE CET APPAREIL DOIVENT SE FAIRESUR ORDONNANCE MÉDICALE OU PAR UN
PRESTATAIRE DE SOINS MUNI D’UNE LICENCE.
AVERTISSEMENT : CET APPAREIL NE DOIT PAS
ÊTRE UTILISÉ À DES FINS DE RÉANIMATION. IL
PEUT ÊTRE NÉCESSAIRE DE SURVEILLER TOUTPARTICULIÈREMENT LES UTILISATEURS N'ÉTANTPAS EN MESURE DE COMMUNIQUER EN CAS DE
MALAISE LORS DE L'UTILISATION DE L'APPAREIL.
LES UTILISATEURS SOUFFRANT DE DÉFICIENCE
AUDITIVE ET/OU VISUELLE PEUVENT AVOIR BESOIN
D’UNE ASSISTANCE POUR SURVEILLER LESALARMES.
AVERTISSEMENT : LES RÉGLAGES DU MODE
D’ADMINISTRATION PULSÉ DOIVENT ÊTREDÉTERMINÉS EN FONCTION DES BESOINSINDIVIDUELS DE CHAQUE UTILISATEUR AU REPOS,
DURANT UN EFFORT ET PENDANT LES VOYAGES.

AVERTISSEMENT : CERTAINS EFFORTSRESPIRATOIRES DU PATIENT PEUVENT NE PAS
DÉCLENCHER LA FONCTION DE CONSERVATION.
AVERTISSEMENT : LE FREESTYLE COMFORT N'ESTPAS DESTINÉ À ÊTRE UTILISÉ AVEC UN PATIENTTRACHÉOTOMISÉ.
AVERTISSEMENT : LES FEMMES ENCEINTES OUQUI ALLAITENT NE DOIVENT PAS UTILISER LESACCESSOIRES RECOMMANDÉS DANS CE MANUEL,CEUX-CI POUVANT CONTENIR DES PHTALATES.

Contre-indications d’utilisation
AVERTISSEMENT : DANS CERTAINS CAS,L’UTILISATION D’OXYGÈNE SANS AVISMÉDICAL PEUT S’AVÉRER DANGEREUSE.
CET APPAREIL DOIT ÊTRE UTILISÉUNIQUEMENT SUR PRESCRIPTIONMÉDICALE.
AVERTISSEMENT : CE DISPOSITIF N’EST PAS
DESTINÉ À ÊTRE UTILISÉ DANS OU À PROXIMITÉD’ENVIRONNEMENTS DE RÉSONANCE MAGNÉTIQUE,TELS QUE DES APPAREILS D’IRM.
AVERTISSEMENT : NE PAS UTILISER CET APPAREIL
EN PRÉSENCE D’ANESTHÉSIQUES INFLAMMABLES.
AVERTISSEMENT : COMME TOUT DISPOSITIF
ÉLECTRIQUE, CET APPAREIL PEUT NE PAS
FONCTIONNER EN RAISON D’UNE COUPURE DE
COURANT OU NÉCESSITER D’ÊTRE RÉPARÉ PAR
UN TECHNICIEN QUALIFIÉ. LE CONCENTRATEURD’OXYGÈNE NE PEUT ÊTRE UTILISÉ POUR LESUTILISATEURS QUI PRÉSENTERAIENT DES
RÉACTIONS INDÉSIRABLES DÉCOULANT D’UNE
INTERRUPTION TEMPORAIRE DE L’APPAREIL.
AVERTISSEMENT : L’UTILISATION DE CE
DISPOSITIF N’EST PAS APPROUVÉE DANSTOUS LES ENVIRONNEMENTS. L’EXPOSITION ÀCERTAINES SOURCES ÉLECTROMAGNÉTIQUESPEUT PROVOQUER DES INTERFÉRENCES OU DESPERFORMANCES INVOLONTAIRES DU DISPOSITIF.
EN CAS D’INTERFÉRENCE, L’APPAREIL DOIT ÊTREARRÊTÉ IMMÉDIATEMENT.

AVERTISSEMENT : CE DISPOSITIF EST DESTINÉÀ ÊTRE UTILISÉ DANS LES ÉTABLISSEMENTS DE
SOINS À DOMICILE ET DANS DES ENVIRONNEMENTS
EXTERNES SIMILAIRES.
AVERTISSEMENT : CE DISPOSITIF EST DESTINÉÀ ÊTRE UTILISÉ PAR LE PATIENT RECEVANT UNEOXYGÉNOTHÉRAPIE.

PN MN232-C4 L | Manuel d’utilisation FR - 41

FreeStyle Comfort
Consignes de sécurité
AVERTISSEMENT : AUCUNE MODIFICATION
DE CET ÉQUIPEMENT N’EST AUTORISÉEAVERTISSEMENT : CET APPAREIL DISPENSE UN
OXYGÈNE EXTRÊMEMENT PUR, SUSCEPTIBLE
DE S’ENFLAMMER TRÈS VITE. NE FUMEZ PAS À
PROXIMITÉ ET ÉVITEZ TOUTE FLAMME NUE DANS LA
MÊME PIÈCE QUE (1) CET APPAREIL, OU QUE (2) TOUTAUTRE ACCESSOIRE DE TRANSPORT D’OXYGÈNE. LE
NON-RESPECT DE CET AVERTISSEMENT PEUT AVOIRPOUR CONSÉQUENCES UN INCENDIE, DES DÉGÂTS
MATÉRIELS ET/OU DES BLESSURES GRAVES VOIREMORTELLES.
AVERTISSEMENT : N’UTILISEZ PAS CET APPAREIL
DANS UN ESPACE FERMÉ OU CONFINÉ (PAR
EXEMPLE DANS UNE PETITE VALISE OU UN SAC)
OFFRANT UNE VENTILATION POTENTIELLEMENTINSUFFISANTE. CECI POURRAIT PROVOQUER UNE
SURCHAUFFE DU CONCENTRATEUR D’OXYGÈNE ET
UNE DÉGRADATION DE SES PERFORMANCES.
AVERTISSEMENT : LE CONCENTRATEUR D’OXYGÈNE
DOIT ÊTRE PLACÉ À L’ÉCART DE TOUTE SOURCE DE
FUMÉE, DE POLLUTION OU DE VAPEURS.
AVERTISSEMENT : EN CAS D’INCIDENT GRAVESURVENANT AVEC CET APPAREIL, L’UTILISATEURDOIT IMMÉDIATEMENT SIGNALER L’INCIDENT AU
FOURNISSEUR ET/OU AU FABRICANT. UN INCIDENTGRAVE EST DÉFINI COMME UNE BLESSURE, UNDÉCÈS OU UN RISQUE DE BLESSURE OU DEDÉCÈS SI L’INCIDENT VENAIT À SE REPRODUIRE.
L’UTILISATEUR PEUT ÉGALEMENT SIGNALER
L’INCIDENT À L’AUTORITÉ COMPÉTENTE DU PAYS
OÙ L’INCIDENT S’EST PRODUIT.
AVERTISSEMENT : LE RECOURS À DES
ACCESSOIRES D’ADMINISTRATIOND’OXYGÈNE NON PRÉVUS POUR UNE
UTILISATION AVEC CE CONCENTRATEURD’OXYGÈNE EST SUSCEPTIBLE DE NUIREAUX PERFORMANCES DE L’APPAREIL.
LES RÉFÉRENCES DES ACCESSOIRESRECOMMANDÉS SONT INDIQUÉES DANSCE MANUEL.
AVERTISSEMENT : EN CAS DE CHUTE,
D'ENDOMMAGEMENT OU D'EXPOSITION DU
CONCENTRATEUR D'OXYGÈNE À DE L'EAU,CONTACTEZ LE FOURNISSEUR DE VOTREÉQUIPEMENT AFIN DE PROCÉDER À L'INSPECTIONOU AUX RÉPARATIONS POSSIBLES DE L'APPAREIL.
VEILLEZ À NE PAS UTILISER LE CONCENTRATEURD’OXYGÈNE SI SA FICHE OU SON CORDON
D’ALIMENTATION ÉLECTRIQUE EST ENDOMMAGÉ(E).
AVERTISSEMENT : VEILLEZ À ATTACHERCORRECTEMENT LE CONCENTRATEUR D’OXYGÈNE
À L’AIDE D’UNE CEINTURE OU PAR TOUT AUTRE
MOYEN APPROPRIÉ PENDANT LE TRANSPORT DANSUN VÉHICULE AFIN D’ÉVITER TOUTE BLESSURE OU
DÉTÉRIORATION.
42 - FR PN MN232-C4 L | Manuel d’utilisation

AVERTISSEMENT : LES RÉGLEMENTATIONS DU
DÉPARTEMENT DES TRANSPORTS DES ÉTATS-UNISET DES NATIONS UNIES (ONU) EXIGENT QUE LA
BATTERIE SOIT RETIRÉE DE L’APPAREIL SUR
TOUS LES VOLS INTERNATIONAUX SI CELUI-CI ESTENREGISTRÉ EN TANT QUE BAGAGE. LORS DE
L’EXPÉDITION DU CONCENTRATEUR D’OXYGÈNE,IL EST ÉGALEMENT IMPÉRATIF DE RETIRER LABATTERIE DE L’APPAREIL ET DE L’EMBALLER
CORRECTEMENT.
AVERTISSEMENT : LISEZ ET ASSUREZ-VOUS D’AVOIRCOMPRIS LES INFORMATIONS DE CE MANUEL AVANT
D’UTILISER CET APPAREIL. SI VOUS NE COMPRENEZ
PAS LES AVERTISSEMENTS ET LES INSTRUCTIONS
DE CE MANUEL, VEUILLEZ CONTACTER VOTRE
FOURNISSEUR AGRÉÉ AVANT D’ESSAYER D’UTILISERL’APPAREIL AFIN D’ÉVITER TOUT RISQUE DE
BLESSURE OU DE DÉGÂTS MATÉRIELS.
AVERTISSEMENT : SI VOUS VOUS SENTEZ MAL
À L’AISE OU EN CAS D’URGENCE MÉDICALE,SOLLICITEZ IMMÉDIATEMENT UNE ASSISTANCEMÉDICALE.
AVERTISSEMENT : L’UTILISATION DU
CONCENTRATEUR D’OXYGÈNE À UNE TEMPÉRATURENON CONFORME AUX CARACTÉRISTIQUESTECHNIQUES PEUT LIMITER SA CAPACITÉ ÀDÉLIVRER LA CONCENTRATION EN OXYGÈNESPÉCIFIÉE. REPORTEZ-VOUS À LA SECTIONCARACTÉRISTIQUES TECHNIQUES DE CE MANUELPOUR CONSULTER LES LIMITES DE TEMPÉRATURE.
AVERTISSEMENT : VEILLEZ À NE PAS UTILISERD’HUILE, DE GRAISSE, DE PRODUITS À BASE
DE PÉTROLE OU D’AUTRES SUBSTANCESINFLAMMABLES AVEC LES ACCESSOIRESDE TRANSPORT D’OXYGÈNE OU AVEC LE
CONCENTRATEUR D’OXYGÈNE. SEULES DESLOTIONS ET DES CRÈMES À BASE D’EAU ET
COMPATIBLES AVEC LA PRÉSENCE D’OXYGÈNE
DOIVENT ÊTRE UTILISÉES. L’OXYGÈNE ACCÉLÈRE LACOMBUSTION DES SUBSTANCES INFLAMMABLES.
AVERTISSEMENT : EN CAS D’UTILISATIONINCORRECTE DE LA BATTERIE, CELLE-CI PEUT
SURCHAUFFER, PRENDRE FEU ET PROVOQUER DESBLESSURES GRAVES. VEILLEZ À NE JAMAIS PERCER,
COGNER OU FAIRE TOMBER LA BATTERIE, MARCHERDESSUS OU L’EXPOSER À DES IMPACTS OU DES
CHOCS VIOLENTS. L’UTILISATION D’UNE BATTERIEENDOMMAGÉE PEUT PROVOQUER DES BLESSURES.
AVERTISSEMENT : VEILLEZ À NE PAS
COURT-CIRCUITER LES CONTACTS EN MÉTAL DE
LA BATTERIE AVEC DES OBJETS MÉTALLIQUES,TELS QU’UNE CLÉ OU UNE PIÈCE DE MONNAIE,
CECI POURRAIT PROVOQUER DES ÉTINCELLESOU LA GÉNÉRATION DE CHALEUR EXCESSIVE.

FreeStyle Comfort
AVERTISSEMENT : SI VOUS UTILISEZ LE
CONCENTRATEUR D’OXYGÈNE PORTABLE À
L’EXTÉRIEUR À L’AIDE D’UNE ALIMENTATION CA,
BRANCHEZ L’ALIMENTATION DANS UNE PRISE AVECDISJONCTEUR DE FUITE DE TERRE UNIQUEMENT.

AVERTISSEMENT : NE TENTEZ PAS
DE RÉALISER DES OPÉRATIONS DE
MAINTENANCE AUTRES QUE LESSOLUTIONS DE DÉPANNAGE DÉTAILLÉESDANS CE MANUEL.

AVERTISSEMENT : LE CONCENTRATEUR D’OXYGÈNE
NE DOIT PAS ÊTRE UTILISÉ LORSQU’IL EST ACCOLÉÀ, OU EMPILÉ AVEC D’AUTRES ÉQUIPEMENTS. SI UN
TEL POSITIONNEMENT NE PEUT PAS ÊTRE ÉVITÉ,IL CONVIENT DE SURVEILLER L’APPAREIL POURS’ASSURER QU’IL FONCTIONNE CORRECTEMENT.

AVERTISSEMENT : UTILISEZ EXCLUSIVEMENT LESACCESSOIRES RECOMMANDÉS PAR LE FABRICANT.
L’UTILISATION D’AUTRES ACCESSOIRES POURRAITS’AVÉRER DANGEREUSE ET ENDOMMAGERGRAVEMENT VOTRE CONCENTRATEUR D’OXYGÈNE,ET PROVOQUERAIT L’ANNULATION DE LA GARANTIE.

AVERTISSEMENT : L’UTILISATION DE CÂBLES ET
D’ADAPTATEURS AUTRES QUE CEUX SPÉCIFIÉS,
À L’EXCEPTION DES CÂBLES ET ADAPTATEURSCOMMERCIALISÉS PAR LE FABRICANT DE
L’ÉQUIPEMENT ÉLECTRO-MÉDICAL POUR LE
REMPLACEMENT DE COMPOSANTS INTERNES,
RISQUE D’ENTRAÎNER UNE AUGMENTATION DESÉMISSIONS ÉLECTROMAGNÉTIQUES OU UNEDIMINUTION DE L’IMMUNITÉ ÉLECTROMAGNÉTIQUEDU CONCENTRATEUR D’OXYGÈNE.

AVERTISSEMENT : LE FABRICANT RECOMMANDE DEDISPOSER D’UNE AUTRE SOURCE D’ALIMENTATIONEN OXYGÈNE EN CAS DE COUPURE DE COURANT, DE
DÉCLENCHEMENT D’ALARME DE L’APPAREIL OU DEPANNE MÉCANIQUE. CONSULTEZ VOTRE MÉDECINOU VOTRE FOURNISSEUR AGRÉÉ AU SUJET DU TYPEDE SYSTÈME DE RÉSERVE À UTILISER.

AVERTISSEMENT : UTILISEZ L’APPAREIL
UNIQUEMENT À LA TENSION ÉLECTRIQUE INDIQUÉE
SUR L’ÉTIQUETTE DES CARACTÉRISTIQUESTECHNIQUES APPOSÉE SUR L’APPAREIL.
AVERTISSEMENT : N’UTILISEZ PAS DE RALLONGE
ÉLECTRIQUE AVEC CET APPAREIL ET VEILLEZ
À NE PAS BRANCHER UN TROP GRANDNOMBRE D’APPAREILS SUR LA MÊME PRISESECTEUR. L’UTILISATION D’UNE RALLONGE
RISQUE D’AFFECTER LES PERFORMANCES DE
L’APPAREIL. DE PLUS, SI VOUS EFFECTUEZ TROP
DE BRANCHEMENTS SUR UNE MÊME PRISE, VOUS
RISQUEZ DE PROVOQUER UNE SURCHARGE DUTABLEAU ÉLECTRIQUE ET DE DÉCLENCHER LEDISJONCTEUR OU DE PROVOQUER UN INCENDIE SI
CELUI-CI NE FONCTIONNE PAS.
MISE EN GARDE : Si une alarme sedéclenche ou si vous constatez que leconcentrateur d’oxygène ne fonctionne pas
correctement, reportez-vous au Tableau dedépannage figurant dans ce manuel. Si vous
ne parvenez pas à résoudre le problème,
consultez votre fournisseur agréé.

AVERTISSEMENT : NE LAISSEZ PAS LESORIFICES D’ADMISSION ET D’ÉVACUATION DE
L’AIR S’OBSTRUER. NE FAITES PAS TOMBERL’APPAREIL OU N'INSÉREZ AUCUN OBJET DANS SESOUVERTURES. CECI POURRAIT PROVOQUER UNE
SURCHAUFFE DU CONCENTRATEUR D’OXYGÈNE ET
UNE DÉGRADATION DE SES PERFORMANCES.
AVERTISSEMENT : FUMER EN UTILISANT DEL’OXYGÈNE CONSTITUE LA PREMIÈRE CAUSE DE
BLESSURES ET DE DÉCÈS LIÉS AUX INCENDIES.
VOUS DEVEZ IMPÉRATIVEMENT RESPECTER LESAVERTISSEMENTS DE SÉCURITÉ SUIVANTS :AVERTISSEMENT : NE LAISSEZ PERSONNE FUMERET N’UTILISEZ PAS DE BOUGIES OU DE FLAMMESNUES DANS LA MÊME PIÈCE QUE L’APPAREILOU QUE LES ACCESSOIRES DE TRANSPORTD’OXYGÈNE.
AVERTISSEMENT : FUMER LORS DU PORT D’UNE
CANULE À OXYGÈNE PEUT PROVOQUER DESBRÛLURES AU VISAGE OU MÊME LA MORT.
AVERTISSEMENT : SI VOUS FUMEZ, CES 3 GESTESPEUVENT VOUS SAUVER LA VIE : ARRÊTEZ LE
CONCENTRATEUR D’OXYGÈNE, RETIREZ LA CANULEET QUITTEZ LA PIÈCE DANS LAQUELLE SE TROUVEL’APPAREIL.

MISE EN GARDE : suivez toutes les instructions
du fabricant afin d’éviter l’annulation de la garantie.

PN MN232-C4 L | Manuel d’utilisation FR - 43

FreeStyle Comfort
AVERTISSEMENT : DES INDICATIONS« INTERDICTION DE FUMER - UTILISATIOND’OXYGÈNE » DOIVENT ÊTRE AFFICHÉES DEFAÇON BIEN VISIBLE À VOTRE DOMICILE OU DANSTOUT LIEU OÙ DE L’OXYGÈNE EST UTILISÉ. LESUTILISATEURS ET LE PERSONNEL SOIGNANT
DOIVENT ÊTRE INFORMÉS DES DANGERS LIÉSAU FAIT DE FUMER EN PRÉSENCE OU PENDANTL’UTILISATION D’OXYGÈNE MÉDICAL.
AVERTISSEMENT : N’UTILISEZ PAS VOTRE
CONCENTRATEUR D’OXYGÈNE EN PRÉSENCEDE GAZ INFLAMMABLES. CELA PEUT PROVOQUERUN EMBRASEMENT RAPIDE SUSCEPTIBLE
D’ENTRAÎNER DES DÉGÂTS MATÉRIELS, DESBLESSURES OU LA MORT.
AVERTISSEMENT : LE FAIT DE RETIRER LA CANULEET DE LA PLACER SUR UN VÊTEMENT, UN DRAP,
UN SOFA OU TOUTE AUTRE MATIÈRE TEXTILE PEUTPROVOQUER UN EMBRASEMENT INSTANTANÉEN CAS D’EXPOSITION À UNE CIGARETTE, À UNE
SOURCE DE CHALEUR OU À UNE FLAMME.
AVERTISSEMENT : NE LAISSEZ JAMAIS UNE CANULE
NASALE SUR OU SOUS UN VÊTEMENT, UN DRAPOU UN COUSSIN. SI L’APPAREIL EST ALLUMÉSANS ÊTRE UTILISÉ, CES MATIÈRES RISQUENTDE S’ENFLAMMER AU CONTACT DE L’OXYGÈNE.
VEILLEZ À ÉTEINDRE LE CONCENTRATEURD’OXYGÈNE LORSQU’IL N’EST PAS UTILISÉ.
AVERTISSEMENT : LES ACCESSOIRESD'ALIMENTATION EN OXYGÈNE APPROUVÉSSONT CONÇUS POUR RÉDUIRE LA PROPAGATION
DE L'INCENDIE POUR LA SÉCURITÉ DU PATIENT.
L’UTILISATION D’ACCESSOIRES NON APPROUVÉSPEUT AUGMENTER LE RISQUE DE BLESSURES OU
DE DOMMAGES LIÉS AUX INCENDIES.
AVERTISSEMENT : LE CONCENTRATEUR D'OXYGÈNE
DOIT ÊTRE ARRÊTÉ ET DÉBRANCHÉ DE TOUTE
SOURCE D'ALIMENTATION EXTERNE AVANT DE
RETIRER LA BATTERIE.
AVERTISSEMENT : LES PETITES PIÈCESPEUVENT PRÉSENTER UN RISQUEPOTENTIEL D'ÉTOUFFEMENT.
AVERTISSEMENT : ÉVITEZ TOUT CONTACTAVEC L'ÉVACUATION D'AIR PENDANT PLUS
DE 10 SECONDES, CAR CELA PEUT ÊTRE CHAUD.
AVERTISSEMENT : LE VENT OU LES FORTS COURANTS
D’AIR PEUVENT NUIRE À L’ADMINISTRATION PRÉCISE
DE L’OXYGÉNOTHÉRAPIE.
AVERTISSEMENT : CE PRODUIT PEUT VOUSEXPOSER À DES PRODUITS CHIMIQUES, Y COMPRISLE NICKEL, QUI EST RECONNU PAR L’ÉTAT DE
CALIFORNIE COMME ÉTANT CANCÉRIGÈNE. POUR
PLUS D’INFORMATIONS, VEUILLEZ VISITER LE SITE
WWW.P65WARNINGS.CA.GOV.

44 - FR PN MN232-C4 L | Manuel d’utilisation

MISE EN GARDE : débranchez toujours
l’alimentation CA de la prise secteur murale
avant de déconnecter l’alimentation CA duconcentrateur d’oxygène.
MISE EN GARDE : ne placez pas le concentrateurd'oxygène de façon à ce qu'il soit difficile d'accéder au
cordon d'alimentation.
MISE EN GARDE : placez toujours les cordonsd’alimentation et la tubulure d’alimentation en oxygènede sorte à empêcher toute personne de trébucher ou
de s’étrangler accidentellement.
MISE EN GARDE : lorsque le concentrateurd’oxygène est utilisé dans une voiture, sur un bateau
ou sur d’autres sources de courant continu avec
l’alimentation CC, assurez-vous que le véhicule
est en marche avant de connecter l’appareil. Si ces
instructions ne sont pas respectées, le concentrateurd’oxygène risque de ne pas être alimenté.
MISE EN GARDE : lorsque le moteur du véhicule danslequel vous utilisez le concentrateur d’oxygène estarrêté, déconnectez l’appareil et retirez-le du véhicule.
Ne stockez pas le concentrateur d’oxygène dans unvéhicule dont l’habitacle peut devenir très chaudou très froid, ou dans tout environnement similaire
pouvant présenter des températures particulièrement
faibles ou élevées.
Remarque : les équipements portables et mobiles decommunication par radiofréquences peuvent affecter lefonctionnement des équipements médicaux électriques.
MISE EN GARDE : VEILLEZ À NE PAS
laisser le concentrateur d’oxygène ou
l’alimentation connecté(e) dans le véhicule
si le moteur est arrêté. afin de ne pas
décharger la batterie du véhicule.
MISE EN GARDE : l’utilisation ou le stockage duconcentrateur d’oxygène en dehors de sa plage detempérature de fonctionnement normale et de laplage de pression ambiante (altitude) peut affecter lesperformances et réduire le temps de fonctionnement
de la batterie et/ou augmenter sa durée de chargement.
Reportez-vous à la section Caractéristiques techniquesde ce manuel pour consulter les limites de température
relatives au stockage et au fonctionnement de l’appareil.
MISE EN GARDE : Pour les concentrateurs d’oxygèneéquipés d’une batterie : afin de préserver la longévité
de votre batterie, stockez-la dans un endroit frais etsec. Le stockage du concentrateur d’oxygène pendantde longues périodes à de hautes températures, avec
une batterie totalement chargée ou entièrementépuisée, peut écourter la durée de vie de celle-ci. Ne
tentez pas d’ouvrir la batterie. Elle ne contient aucune
pièce à entretenir ou à remplacer. Veillez à tenir labatterie hors de portée des enfants.
MISE EN GARDE : UTILISEZ UNIQUEMENT les batteries
fournies par le fabricant. Contactez le fournisseurde l’équipement ou votre mairie pour connaître les
obligations en matière de recyclage des batteries.

FreeStyle Comfort
Remarque : Pour atténuer les dommages causés par leconcentrateur ou une panne de courant, le fournisseurd'équipement doit évaluer la nécessité d'oxygène desecours pendant l'installation et périodiquement lorsqueces attributs changent :- l’état du patient- l’environnement dans lequel vit le patient- la possibilité de réalimenter le patient en oxygène
supplémentaire.
Remarque : vérifiez que la canule est entièrement
insérée et convenablement positionnée. Le concentrateurd’oxygène peut ainsi détecter correctement chaqueinspiration pour délivrer l’oxygène requis. Durant
l’inhalation, vous devez entendre ou sentir l’oxygène
circuler dans les sondes de la canule nasale.

Remarque : si le concentrateur d’oxygène a été stocké
pendant une période prolongée dans un environnement
ne respectant pas les conditions de fonctionnement
normales décrites dans le tableau ci-dessous, il doit
tout d’abord être placé dans un endroit lui permettant
d’atteindre une température de fonctionnement normale
avant de pouvoir être allumé.
Remarque : le fournisseur d'équipement doit évaluer lanécessité d'oxygène de secours pendant l'installation etpériodiquement lorsque ces attributs changent :- l’état du patient- l’environnement dans lequel vit le patient- la possibilité de réalimenter le patient en oxygène
supplémentaire.

Remarque : Les batteries au lithium peuvent neplus fonctionner lorsqu’elles sont exposées à destempératures extrêmement élevées alors qu’elles sonttotalement chargées ou entièrement déchargées.
Pour un stockage prolongé, il est recommandé decharger les batteries entre 25 et 50 % et de les conserverà une température de -20 °C à 20 °C (-4 °F à 68 °F).
Remarque : n’utilisez pas le concentrateur d’oxygène
lorsque le filtre d’admission d’air n’est pas en place.
Si un second filtre est fourni, insérez le filtre deremplacement avant de nettoyer le filtre sale. Nettoyez lefiltre sale dans de l’eau chaude savonneuse, puis
laissez-le sécher totalement avant de l’utiliser.
Remarque : le fabricant ne recommande pas de stérilisercet équipement.
Remarque : l’utilisation du concentrateur d’oxygène nerisque en aucun cas d’épuiser tout l’oxygène de la pièce.
AVERTISSEMENT : LE CONCENTRATEURD'OXYGÈNE NE DOIT PAS ÊTRE UTILISÉOU BRANCHÉ DANS UNE ALIMENTATIONEXTERNE (AC/DC) SANS QU’AUCUNEBATTERIE NE SOIT INSTALLÉE. NE
TENTEZ PAS DE METTRE EN MARCHE
L’APPAREIL OU DE LE BRANCHER À UNE
ALIMENTATION EXTERNE LORSQUE LA
BATTERIE EST RETIRÉE.
RETIREZ UNIQUEMENT LA BATTERIE LORSQUEL’APPAREIL EST ÉTEINT ET DÉBRANCHÉ DEL’ALIMENTATION. NE TENTEZ PAS DE RETIRER LABATTERIE LORSQUE L’APPAREIL EST EN MARCHEOU CONNECTÉ À L’ALIMENTATION.

PN MN232-C4 L | Manuel d’utilisation FR - 45

FreeStyle Comfort
Caractéristiques techniques
FreeStyle Comfort
Dimensions (avec une seule batterie)

10,0 x 7,3 x 3,1 in (25,4 x 18,5 x 7,9 cm)

Dimensions (avec double batterie)

11,0 x 7,3 x 3,1 in (27,9 x 18,5 x 7,9 cm)

Poids (avec batterie simple)

2,3 kg (5 lb)

Poids (avec double batterie)

2,7 kg (6 lb)

Niveau sonore nominal***39,93 dB(A) ± 0,42 dB(A) à 2

Réglages du débit

Réglages pulsés 1, 2, 3, 4, 5

Paramètre de débit d’O2

Paramètre 1 – 210 ml/min
Paramètre 2 – 420 ml/min
Paramètre 3 – 630 ml/min
Paramètre 4 – 840 ml/min
Paramètre 5 – 1050 ml/min([à20bpm STPD] Tous les volumes +/- 15 %)Concentration en oxygène*
Moniteur d’oxygène

90 % (+5,5 % / -3 %)OuiPression de sortie maximum<30 PSI

Sensibilité de déclenchement> -0,1 cm H2O (réglable)

Alimentation CA

100 à 240 V CA, 50 à 60 Hz, 120 W max

Alimentation CC

11–18 VDC (10 amp max)Type de batterie

Lithium-ion

Capacité de la batterie
Temps de rechargement de la batterie
Autonomie de la batterie
Température de fonctionnement**

Cellule unique : 6700 mAh ; cellule double : 13 400 mAh
Une batterie : 3 h 30
Batterie double : 6 hDurée de la batterie simple : Paramètre 2 – jusqu'à 4 heures
Durée de la batterie double : Paramètre 2 – jusqu'à 8 heures5 à 40°C (41 °F à 104 °F)

Température de transport et de stockage (Unité)-25°C à 70°C (-13°F à +158°F) jusqu'à 3 048 m (10 000 pieds)

Température de transport et de stockage(batterie)-25°C à 70°C (-13°F à +158°F) jusqu'à 3 048 m (10 000 pieds)

Altitude de fonctionnement**-381 à 3 048 m (-1 250 à 10 000 pieds) (Testé de 700 à 1 060 hPa)

Humidité de fonctionnement

15 – 95 % d’humidité relative (sans condensation)

Humidité de stockage

0 à 90 % sans condensation

Garantie limitéeVoir déclaration de garantie* À une pression atmosphérique de 101 kPa (14,7 psi) et une température de 21°C (70°F).
** L'appareil fonctionnera dans tous les environnements tant que l'environnement est dans les plages de température
et d'humidité spécifiées. Le fonctionnement en dehors de ces caractéristiques techniques peut limiter la capacité duconcentrateur à fournir une concentration en oxygène adéquate à des débits plus élevés. L'écran de préchauffage peut
continuer à s'afficher jusqu'à ce que le concentrateur ait atteint la concentration cible ou jusqu'à 15 minutes. Continuez
d'utiliser l'appareil pendant ce temps.
*** Niveau sonore mesuré par méthode de test N° 14-1 10/2018 MDS-Hi.
La durée de vie estimée est d’au moins cinq ans.

46 - FR PN MN232-C4 L | Manuel d’utilisation

FreeStyle Comfort
Instructions d’utilisation
1. Localisez et placez le FreeStyle Comfort dans un
endroit approprié.
a. Le concentrateur doit être correctement
ventilé, de sorte que les orifices d’admission etd’évacuation d’air ne soient pas obstrués.
b. Le concentrateur doit être placé de façon à ceque tous les indicateurs et alarmes sonores etvisuels puissent être facilement vus et entendus.
c. Veillez à ce que les filtres d’admission d’air
soient placés avant d’utiliser l’appareil. Si un
filtre de l’appareil est manquant, contactez lefournisseur de votre équipement.
2. Placez la batterie dans le concentrateur.
a. Alignez la batterie sur le logement inférieur
depuis le centre du FreeStyle Comfort.

3. Insérez le bloc d’alimentation CA ou CC dans leréceptacle du port d’alimentation sur l’avant, en basà droite de l’appareil.

4. Connectez une canule nasale au raccord de sortie del’oxygène. Veillez à ce que le raccord soit solidement
fixé avant d’utiliser l’appareil.
REMARQUE : L’utilisation de toute canule autre que cellerecommandée peut affecter les performances.

b. Faites glisser la batterie à l’intérieur jusqu’à
entendre le clic confirmant sa bonne position.

Remarque : À l’attention du fournisseur de l’équipement :les accessoires d’administration d’oxygène recommandésavec le FreeStyle Comfort sont les suivants :• Canule nasale et tubulure de 2,1 m (7 pieds) de long
(6 l/min) : CAIRE Référence 5408-SEQ• Des informations supplémentaires sur les accessoires
recommandés sont disponibles en ligne sur www.caireinc.
com.
5. Allumez le FreeStyle Comfort en appuyant sur lebouton d'alimentation et sélectionnez le réglage dedébit qui vous a été prescrit en utilisant les touches
d’augmentation (+) ou de diminution (-) du réglage
de débit.

PN MN232-C4 L | Manuel d’utilisation FR - 47

FreeStyle Comfort

Une brève alarme sonore retentira pour indiquer que
l'appareil est sous tension.
Remarque : Lorsque le FreeStyle Comfort est mis en
marche, le paramètre de débit par défaut est 2. Si leréglage est modifié, l’appareil s’allume sur le réglage dedébit précédent.
6. Placez la canule nasale sur votre visage
conformément aux instructions du fabricant fournies
avec la canule, puis respirez normalement. Vous
devriez ressentir une bouffée d’oxygène provenantde la canule nasale à chaque fois que vous inspirez.
7. Pour modifier le réglage du débit, appuyez sur les
touches d’augmentation (+) ou de diminution (-) duréglage de débit pour obtenir le débit souhaité. Il est
normal qu’une différence de son se fasse entendre
lorsque vous modifiez les réglages du débit.

Taux derespiration
15
20
25
30
35
40

1

Volume d’impulsions non
autoSat (ml)* Réglages
2
3
4

5

14,0
10,5
8,4
7,0
6,0
5,3

28,0
21,0
16,8
14,0
12,0
10,5

70,0
52,5
42,0
35,0
30,0
26,3

42,0
31,5
25,2
21,0
18,0
15,8

56,0
42,0
33,6
28,0
24,0
21,0

8. Éteignez le FreeStyle Comfort en appuyant sur lebouton d’alimentation.
Remarque : L'écran d'affichage passe en mode veille etdiminue après 20 secondes. Appuyez sur n’importe quel
bouton pour sortir l’écran du mode veille.
9. Pour découvrir les fonctionnalités avancées,consultez les instructions de navigation du menu

PARTICULARITÉS DU PRODUIT
autoSAT®
La fonction autoSAT exclusive de CAIRE permet au
FreeStyle Comfort de délivrer un volume constantd’oxygène (bolus) par respiration, jusqu’à un rythme
respiratoire maximal. Lorsque le rythme respiratoirede l’utilisateur augmente lors d’activités telles que lamarche, la fonction autoSAT ajuste le fonctionnement
de l’appareil pour maintenir un volume de bolus défini.
UltraSense®
Ce déclencheur de dose pulsée sensible détectele changement de pression créé par votre respiration
et libère une dose d’oxygène lorsque vous inspirez.

Taux derespiration
15
20
25
30
35
40

1

Volume d’impulsions
autoSat (ml)*Réglages
2
3
4

5

10,5
10,5
10,5
10,5
10,5
10,5

21,0
21,0
21,0
21,0
21,0
21,0

52,5
52,5
42,0
35,0
30,0
26,3

31,5
31,5
31,5
31,5
30,0
26,3

42,0
42,0
42,0
35,0
30,0
26,3

*La tolérance est +/- 15 %48 - FR PN MN232-C4 L | Manuel d’utilisation

autoDOSE®
En cas d’alarme de type « Aucun souffle détecté »,cette fonction entraînera la sortie d’oxygène à raisonde 20 respirations par minute selon les réglagesdu débit actuels jusqu’à ce que la cause de l’alarmesoit réglée.

FreeStyle Comfort
Alimentations
Le FreeStyle Comfort peut être alimenté de trois façons :• Bloc-batterie rechargeable• Alimentation CA• Alimentation CC
Les informations relatives au fonctionnement de ces trois
composants sont reprises dans les sections suivantes.
AVERTISSEMENT : UTILISEZ UNIQUEMENTLES ALIMENTATIONS DÉCRITES DANS CE
MANUEL D’UTILISATION. L’UTILISATION
DE BATTERIES ET D’ALIMENTATIONSAUTRES QUE CAIRE PEUT ENDOMMAGER L’APPAREIL ET PERTURBER SON
FONCTIONNEMENT.

RETRAIT DE LA BATTERIE

Poussez le bouton de dégagement de la batterie versle bas pour la sortir du FreeStyle Comfort. Tout en
MAINTENANT ce bouton, faites glisser la batteriepour la faire sortir du concentrateur.
AVERTISSEMENT : LE CONCENTRATEURD'OXYGÈNE DOIT ÊTRE ARRÊTÉ ET
DÉBRANCHÉ DE TOUTE SOURCE
D’ALIMENTATION EXTERNE AVANT
DE RETIRER LA BATTERIE.

BATTERIELe FreeStyle Comfort peut fonctionner avec la batterierechargeable fournie avec l’équipement. Aucune limite
de fonctionnement ne s’applique au concentrateur
lorsqu’il fonctionne sur batterie. Tous les débits sontégalement disponibles.

INSTALLATION DE LA BATTERIE

Alignez la batterie sur le logement inférieur depuis
le centre du FreeStyle Comfort. Poussez la batterie à
l’intérieur jusqu’à entendre le clic confirmant sa bonne
position.

VÉRIFICATION DE LA CHARGE
DE LA BATTERIE

Pour vérifier la charge de la batterie lors de son
installation dans le FreeStyle Comfort, reportez-vousà l'icône de la batterie et à la durée estimée restanteaffichée sur l'écran.
80–100 %60–80 %

40–60 %

20–40 %20 %PN MN232-C4 L | Manuel d’utilisation FR - 49

FreeStyle Comfort
Remarque : si l'appareil est alimenté en courant, un
indicateur de prise s'affiche à la place du pourcentage
de charge restant.
Pour vérifier la charge de la batterie lorsqu’elle n’est
pas installée dans l’appareil FreeStyle Comfort, posez-laà plat et appuyez sur le bouton TEST. Situés à droite
du bouton TEST, les témoins lumineux de l’indicateurde la batterie (25–100 %) s’allument pour indiquer leniveau de charge de la batterie.

DURÉE D'AUTONOMIE DE LA BATTERIE

Lorsque l'alimentation CA ou CC externe est
débranchée, le concentrateur bascule automatiquement
sur la batterie si une batterie chargée est installée.
Le tableau ci-après présente l’autonomie généralement
affichée par une nouvelle batterie. L’autonomie d’une
batterie reste constante indépendamment de votre
fréquence respiratoire.
Réglageintermittent

Durée de la batterie
simple (8 cellules)

Durée de la batterie
double (16 cellules)

1jusqu’à 8 heuresjusqu’à 16 heures2

jusqu’à 4 heuresjusqu’à 8 heures3

jusqu’à 3 heuresjusqu’à 6 heures4

jusqu’à 2,25 heuresjusqu’à 4,5 heures5

jusqu’à 2 heuresjusqu’à 4 heures

DURÉE TYPIQUE DE RECHARGEMENTDE LA BATTERIEPREMIÈRE CHARGE

La nouvelle batterie fournie avec votre FreeStyle
Comfort n'est que partiellement chargée lorsqu'elle est
expédiée à sa sortie d'usine. Avant de l’utiliser pour lapremière fois, vous devez la charger complètement àl’aide d’une alimentation CA ou CC.

INFORMATIONS GÉNÉRALES DE
CHARGEMENTLe FreeStyle Comfort recharge la batterie à chaquefois qu'elle est installée dans le concentrateur et que leconcentrateur est connecté à une source d'alimentation
CA ou CC externe (sauf dans un avion).
La batterie est complètement chargée lorsqu'elle
affiche 100 %.
La batterie peut également être chargée à l’aide duchargeur de bureau fourni en option.

50 - FR PN MN232-C4 L | Manuel d’utilisation

La durée typique pour recharger complètement votre
batterie à partir d'une batterie complètement déchargée
est de 3,5 heures pour une batterie simple et de6 heures pour une batterie double.
Remarque : Il est possible que la batterie ne commence pasà se charger si la batterie est complètement déchargée.

DURÉE DE VIE DE LA BATTERIE

Les batteries FreeStyle Comfort sont conçues pourdurer 300 cycles de charge/décharge.

FreeStyle Comfort
Alimentation CA et CC
Le FreeStyle Comfort est livré standard avec dessecteurs d’alimentation CA et CC.
L’alimentation CA inclut les composants suivants :	a. Transformateur CA qui fonctionne entre
100–240 volts, 50–60 Hz.
	b. Cordon d’alimentation CA pour laconnexion au transformateur et à la prise
CA correspondante.

L’alimentation CC consiste en un cordon qui se
branche directement à l’appareil et à la prise CC.

2. Branchez le cordon d’alimentation CA autransformateur.

ALIMENTATION CA

L’alimentation CA est destinée à être utilisée au
domicile ou à tout endroit où une prise CA standard
est disponible. Le FreeStyle Comfort fonctionne
parfaitement sur une alimentation CA. Le FreeStyle
Comfort recharge également la batterie chaque fois quel'alimentation CA est disponible (sauf dans un avion).
La batterie se recharge, que l’appareil FreeStyle
Comfort soit allumé ou non.
AVERTISSEMENT : NE BRANCHEZ
PAS L’ALIMENTATION EXTERNE SI LABATTERIE N’EST PAS INSTALLÉE DANSLE CONCENTRATEUR D'OXYGÈNE.
Pour brancher l’appareil FreeStyle Comfort sur une
alimentation CA :1. Branchez le transformateur CA à l'entrée duconnecteur d'alimentation à l'avant, en bas à droite
du FreeStyle Comfort. Veillez à bien l’enfoncer.

3. Raccordez l’extrémité opposée du cordon
d’alimentation CA à une prise murale ou à une priseélectrique appropriée.

PN MN232-C4 L | Manuel d’utilisation FR - 51

FreeStyle Comfort
4. Allumez le FreeStyle Comfort en appuyant sur lebouton d'alimentation et sélectionnez le réglage dedébit qui vous a été prescrit en utilisant les touches
d’augmentation (+) ou de diminution (-) du réglage
de débit.

4. Allumez le FreeStyle Comfort en appuyant sur lebouton d'alimentation et sélectionnez le réglage dedébit qui vous a été prescrit en utilisant les touches
d’augmentation (+) ou de diminution (-) du réglage
de débit.
Remarque : si la batterie doit être rechargée, l'icône de labatterie sur l'écran aura une seule barre orange.

ALIMENTATION CC

L’alimentation CC est destinée à être utilisée dans lesprises pour accessoires installées dans les véhicules, lesbateaux et tout autre véhicule à moteur. Le FreeStyle
Comfort est entièrement fonctionnel pour une utilisationen courant continu (CC). Si nécessaire, le FreeStyle
Comfort permet également de recharger la batterielorsqu’une alimentation CC est disponible. La batterie
se recharge si l'appareil est éteint, allumé et en coursd'utilisation.
AVERTISSEMENT : NE BRANCHEZ
PAS L'ALIMENTATION EXTERNE SI LABATTERIE N'EST PAS INSTALLÉE DANSLE CONCENTRATEUR D'OXYGÈNE.
1. Démarrez votre véhicule.
2. Branchez l'alimentation CC à l'avant du réceptacled'alimentation, en bas à droite du FreeStyle
Comfort. Veillez à bien l’enfoncer.

3. Raccordez l’extrémité opposée du cordon
d’alimentation CC à un allume-cigare ou à une
prise CC pour accessoire.

52 - FR PN MN232-C4 L | Manuel d’utilisation

MISE EN GARDE : Pour éviter tout risqued'incendie ou de surchauffe, assurez-vousque la prise d'alimentation 12 V CC est
exempte de débris et que la fiche del'adaptateur de l'alimentation CC s'adapte
correctement et reste fermement en place.
MISE EN GARDE : les surfaces métalliquesd'alimentation en courant continu peuvent devenirTRÈS CHAUDES en cours d'utilisation en raisond'un courant élevé. Laisser refroidir les surfaces
suffisamment longtemps avant de les toucher.
MISE EN GARDE : n'utilisez pas l'alimentation CC avec
des extensions, des séparateurs ou des adaptateurs.
L’utilisation d’accessoires non approuvés peut
provoquer une surchauffe ou des dommages.
MISE EN GARDE : la fixation de l'appareil à une prise decourant 12 V CC peut épuiser la batterie du véhicule si lemoteur ne tourne pas, même si l'appareil est éteint.
MISE EN GARDE : ne démarrez pas un véhicule
lorsque l’alimentation CC est branchée.

FreeStyle Comfort
Nettoyage, entretien
et maintenance régulière
MAINTENANCE RÉGULIÈRE

Maintenance préventive
Le dispositif ne nécessite aucun entretien. Les filtres
à l'intérieur du dispositif sont conçus pour fonctionner
pendant toute la durée de vie du dispositif. Si le dispositif
ne fonctionne pas correctement, l’appareil doit êtreretourné au fournisseur pour réparation. L’entretien descomposants internes du boîtier de l’appareil FreeStyle
Comfort doit être réalisé par votre fournisseur d’oxygène.
L’utilisateur de l'appareil est responsable de laprocédure de maintenance suivante.
• Toutes les semaines – Nettoyer les filtres d’admission
d’air
AVERTISSEMENT : N’ESSAYEZ PAS
D’OUVRIR OU DE RETIRER LES CAPOTSEN PLASTIQUE DU CONCENTRATEURD'OXYGÈNE. L’APPAREIL PEUT
UNIQUEMENT ÊTRE OUVERT PAR UN
TECHNICIEN QUALIFIÉ.

FILTRES D’ADMISSION D’AIR

L’air entre dans l’appareil FreeStyle Comfort par lebiais de deux filtres d’admission d’air situés sur lapartie inférieure avant du concentrateur d’oxygène.
Ces filtres empêchent la poussière et d’autres grandes
particules contenues dans l’air d’entrer dans l’appareil.
Les filtres d’admission d’air doivent être nettoyés au
moins une fois par semaine. En fonction des conditions
d’utilisation, votre fournisseur d’oxygène peut vousconseiller de le nettoyer plus régulièrement. Respectez
ces étapes pour nettoyer correctement les filtresd’admission d’air.
1. Retirez les deux filtres situés à l'avant du FreeStyle
Comfort en appuyant sur l'onglet correspondant et en
tirant vers l'extérieur.

AVERTISSEMENT : N’UTILISEZ PAS LE
CONCENTRATEUR D’OXYGÈNE LORSQUE
LE FILTRE D’ADMISSION D’AIR N’EST PAS
EN PLACE.

MAINTENANCE DE LA BATTERIE
N’exposez pas la batterie à des liquides.

Ne tentez pas de démonter ou d’ouvrir la batterie.
N’exposez pas la batterie aux flammes nues.
Ne laissez pas tomber la batterie.
Maintenez la batterie hors de portée des enfants.
Si votre batterie semble endommagée, contactez votre
fournisseur d’oxygène avant de l’utiliser.
Ne placez pas la batterie dans un four à micro-ondes
ou tout autre appareil de cuisson ni à proximité de cesderniers.

STOCKAGE DE LA BATTERIE

Il n’est pas nécessaire d’attendre le déchargement
complet de la batterie du concentrateur d’oxygène
avant de la recharger, Il est recommandé de charger
la batterie du concentrateur d'oxygène après chaqueutilisation.
Ne laissez pas la batterie dans un endroit exposé à unechaleur excessive, en plein soleil ou dans une voiture,
par exemple, au risque de dégrader les performances
et la durée de vie de la batterie. La batterie pourraitégalement générer de la chaleur et de la fumée, prendre
feu ou exploser. La batterie doit être conservée dansun environnement stable, caractérisé par une faible
humidité (humidité relative inférieure à 70 %) etl’absence de gaz corrosifs. Afin d’empêcher touteapparition de rouille, évitez les endroits susceptibles
de créer de la condensation, notamment où latempérature ambiante peut rapidement varier.
Pour un stockage d’une période égale ou supérieureà un mois, la batterie doit être totalement déchargéeou chargée entre 50 et 75 % (deux des trois témoins
lumineux). Une fois chargée, retirez la batterie duconcentrateur d’oxygène.

2. Nettoyez les deux filtres dans une eau savonneusetiède.
3. Rincez soigneusement les filtres et essuyez l’excèsd’eau à l’aide d’une serviette douce et absorbante.
Veillez à ce que les filtres soient totalement secs
avant de les replacer.
4. Réinstallez les filtres propres et secs.
PN MN232-C4 L | Manuel d’utilisation FR - 53

FreeStyle Comfort
NETTOYAGEArrêtez l’appareil et débranchez-le del’alimentation CA ou CC avant toute opération de
nettoyage. Ne vaporisez PAS de liquides directement
sur le boîtier extérieur. Utilisez une éponge ou un
chiffon légèrement humide. Vaporisez une solution
de nettoyage douce sur l'éponge ou sur le chiffon afinde nettoyer le boîtier et les alimentations. Respectez les
instructions du fabricant du produit de nettoyage.
AVERTISSEMENT : ÉTEIGNEZL’APPAREIL ET DÉBRANCHEZ LE
CORDON D’ALIMENTATION DE LA PRISE
ÉLECTRIQUE AVANT DE NETTOYERL’APPAREIL AFIN D’ÉVITER TOUT CHOC
ÉLECTRIQUE ACCIDENTEL OU RISQUE
DE BRÛLURE. SEUL LE FOURNISSEUR DE
L’APPAREIL OU UN TECHNICIEN QUALIFIÉ
EST HABILITÉ À RETIRER LES CAPOTSOU À RÉPARER L’APPAREIL.
AVERTISSEMENT : PRENEZ TOUTES LESPRÉCAUTIONS POUR ÉVITER DE MOUILLER LE
CONCENTRATEUR D’OXYGÈNE ET LA BATTERIE,ET ASSUREZ-VOUS QU’AUCUN LIQUIDE NE PEUTPÉNÉTRER À L’INTÉRIEUR DE L’APPAREIL AFIN
D’ÉVITER TOUT DYSFONCTIONNEMENT OU ARRÊT
DE CELUI-CI, AINSI QU’UN RISQUE ACCRU DE CHOC
ÉLECTRIQUE OU DE BRÛLURE.
AVERTISSEMENT : NE DÉVERSEZ PAS DE LIQUIDEDIRECTEMENT SUR L’APPAREIL. UNE LISTE
D’AGENTS CHIMIQUES INDÉSIRABLES COMPREND,
SANS TOUTEFOIS S’Y LIMITER, LES SUBSTANCESSUIVANTES : ALCOOL ET PRODUITS À BASE
D’ALCOOL, PRODUITS CONCENTRÉS À BASE DE
CHLORE (CHLORURE D’ÉTHYLÈNE) ET PRODUITSÀ BASE D’HUILE CES PRODUITS NE DOIVENT PAS
ÊTRE UTILISÉS POUR NETTOYER LE BOITIER EN
PLASTIQUE DU CONCENTRATEUR, AU RISQUE
D’ENDOMMAGER L’APPAREIL.
AVERTISSEMENT : NETTOYEZ LE BOÎTIER,
LE PANNEAU DE COMMANDE ET LE CORDON
D’ALIMENTATION UNIQUEMENT AVEC UN CHIFFONOU UNE ÉPONGE HUMIDE (PAS MOUILLÉE) IMBIBÉE
D’UN PRODUIT DE NETTOYAGE DOUX, PUISESSUYEZ TOUTES LES SURFACES. NE LAISSEZAUCUN LIQUIDE PÉNÉTRER À L’INTÉRIEUR DU
CONCENTRATEUR. FAITES PARTICULIÈREMENTATTENTION À LA SORTIE D’OXYGÈNE DESTINÉE
AU RACCORD DE LA CANULE, AFIN QU’ELLE NE
CONTIENNE NI POUSSIÈRE, NI EAU, NI PARTICULES.

54 - FR PN MN232-C4 L | Manuel d’utilisation

AVERTISSEMENT : UTILISEZ UN CHIFFON OU UNEÉPONGE HUMIDE (PAS MOUILLÉE) POUR NETTOYER
LA BATTERIE. IMBIBEZ TOUT D’ABORD LE CHIFFON
OU L’ÉPONGE D’UN PRODUIT DE NETTOYAGE DOUXET NETTOYEZ LE BOÎTIER DE LA BATTERIE.
AVERTISSEMENT : NE PAS NETTOYER AVECUN SYSTÈME DE LAVAGE AUTOMATISÉ.

REMPLACEMENT DE LA CANULE

Suivez toujours les instructions du fabricant de lacanule pour une utilisation correcte. Remplacer lacanule nasale ou la tubulure d’oxygène tous les 3 à
6 mois (si utilisation quotidienne, tous les 3 mois),tel que recommandé par le fabricant de la canule ou
votre fournisseur d’oxygène. Votre médecin ou votre
fournisseur d’oxygène vous transmettra des consignes
concernant son nettoyage et son remplacement.
Des pièces de rechange peuvent également êtreobtenues auprès de votre fournisseur d’oxygène.

FreeStyle Comfort
Conditions d’alarme
N° deprioritéDescription
de l’alarme

1

Dysfonctionnement dusystème Hautetempératurede la batterie

Écran

SignalsonorePlus d’informations/Action suggérée

1 bip
toutes les
20 secondes

L'alarme de la batterie se déclenche lorsquela température de la batterie approche duseuil de sécurité.
Si le seuil de coupure est atteint, l'appareils'éteint. Si l’alimentation externe est
connectée, l’écran d’alarme s’affiche etl’alarme sonore continue pendant 2 minutes.
Effectuez le test avec la batterie à
température ambiante. Si le problème
persiste, contactez le fournisseur.

2

Dysfonctionnement
du systèmeTempératureélevée1 biptoutes les
20 secondes

L'appareil émet une alarme lorsque la température interne approche du seuil de coupure.
Si le seuil de coupure est atteint, l’unités’éteint et l’écran d’alarme s’affiche, puis
l’alarme sonore continue pendant 2 minutes.
Assurez-vous d’éviter toute obstruction desorifices de ventilation de l’appareil. Si leproblème persiste, contactez le fournisseur.

3

Dysfonctionnement dusystème Hautepression (pression élevée dusystème ou
défaillance ducompresseur)

1 biptoutes les
20 secondes

4

Dysfonctionnement
du systèmede débit(défaillance dela valve OCD,
obstruction ducircuit de gaz)1 biptoutes les
30 secondes

Si le problème persiste, contactez lefournisseur.

5Arrêt pourbatterie faible

1 bip
toutes les
30 secondes

Alarmes jusqu’à épuisement de la batterie etl’arrêt de l’appareil.

Si le seuil de coupure est atteint, l’unités’éteint et l’écran d’alarme s’affiche, puis
l’alarme sonore continue pendant 2 minutes.
Si le problème persiste, contactez lefournisseur.

Chargez la batterie.

PN MN232-C4 L | Manuel d’utilisation FR - 55

FreeStyle Comfort
N° deprioritéDescription
de l’alarme

6

Aucun souffle
détectéÉcran

SignalsonoreDétails supplémentaires

1 bip toutes
les 10 min

Aucun souffle détecté pendant 15 secondes.
Activé après le premier souffle ou à l’arrêt del’écran de préchauffage, selon la premièreéventualité.
Vérifiez que la canule n’est pas tordue ou
qu’elle ne fuit pas. Augmentez le réglage de lasensibilité. Si le problème persiste, contactez
le fournisseur.

7

Préchauffage

aucun

L'appareil commence par le dosage
automatique jusqu'à ce qu'un souffle soit
détecté ou que préchauffage soit terminé,selon la première éventualité.

8

Faibleconcentration

1 bip toutes
les 10 min,pendant
jusqu’à
30 minSi le problème persiste, contactez lefournisseur.

9

Alerte debatterie faible

1 bip,
pas derépétition

Chargez la batterie.

10

Dépassementde lafréquence
respiratoire

1 bip,
pas derépétition

Le taux d’inhalation dépasse 40 bpm.

11

Dysfonctionnement
du système
Panne duventilateuraucun

Si le problème persiste, contactez lefournisseur.

12

Dysfonctionnement du système Faible
pressionaucun

Si le problème persiste, contactez lefournisseur.

13

Problème defonctionnement dusystème,température
basse

aucun

Laissez l’appareil se réchauffer à températureambiante. Si le problème persiste, contactez
le fournisseur.

56 - FR PN MN232-C4 L | Manuel d’utilisation

Si la dose est déclenchée de manière
incorrecte, diminuez le réglage de sensibilité.
Si le problème persiste, contactez lefournisseur.

FreeStyle Comfort
N° deprioritéDescription
de l’alarme

Informations

Démarrage

Écran

SignalsonoreDétails supplémentaires

1 bip d'information

L'écran s'affiche à la mise sous tension etvous entendez un bip pour confirmer que lebuzzer fonctionne
Démarrez l'affichage de la révision actuelle dufirmware dans le coin inférieur droit.

Informations

Écran d’exécution normale
en mode autoSAT (aucune
alimentation
CC branchée)aucun

Informations

Écran d’exécution normale
avec autoSAT
désactivé(aucunealimentation
CC branchée)

Aucun

Informations

Le paramètrea été changé

1 bip d'information

Informations

Brancherl'alimentation
externe à
l'arrêt

aucun

Informations

Brancherl'alimentation
externe à l'étatde fonctionnement

aucun

Informations

Débrancher
l'alimentation
externelorsquel'appareil està l'arrêt

1 bip d'information

Affichage de l'écran CAIRE pendant 2
secondes, suivi de l'état de la batterie et de laprise pendant 5 secondes avant que l'écran
s'éteigne

Affichage de l’écran CAIRE pendant2 secondes, puis l’écran s’éteint.

PN MN232-C4 L | Manuel d’utilisation FR - 57

FreeStyle Comfort
N° deprioritéDescription
de l’alarme

Informations

Débrancher
l'alimentation
externelorsque l'appareil fonctionne(avec batterie
présente)

1 bip d'information

Informations

En modeveille - lorsquele bouton
haut ou bas
est enfoncé,
l'affichage
indique l'étatde la batterie
et de charge

aucun

Informations

Mode Avion
lorsquel'appareil est
sur la batterie

1 bip long

Informations

Mode Avion
lorsquel'appareil est
branché

1 bip longÉcran

58 - FR PN MN232-C4 L | Manuel d’utilisation

SignalsonoreDétails supplémentaires

L'écran s'affiche pendant 5 secondes
et s'éteint

FreeStyle Comfort
Instructions de navigation du menu
Vous pouvez accéder au menu en mode veille ou
en mode d’exécution. En mode d’exécution, si
une condition d’alarme survient, vous quitterez
automatiquement le menu. Si vous n’appuyez sur
aucune touche pendant 20 secondes, vous quitterez
automatiquement le menu. En veille, vous devrez
peut-être appuyer deux fois sur le bouton Menu pourréactiver l’appareil, puis accéder au menu.
Il existe quatre écrans de menu :Mode avion

1. Écran d’état : Affiche le nombre de rév. dumicrologiciel, la durée d’exécution, les cycles dedécharge de la batterie
2. Mode avion (MARCHE/ARRÊT)3. Sensibilité : réglable (1–5)
4. autoSAT (MARCHE/ARRÊT)Sensibilité

Exemple d’écran d’état

FONCTIONNEMENT DU MENU RÉGLAGES
autoSAT

Vous pouvez accéder au menu Réglages en appuyant
sur le bouton Menu/Sélectiond’exécution ou veille.

en modesPour faire défiler les menus, appuyez sur le bouton.

Page d’étatPN MN232-C4 L | Manuel d’utilisation FR - 59

FreeStyle Comfort
PAGE D’ÉTAT

Pour annuler le mode Marche du mode avion, appuyez

La page d’état du FreeStyle Comfort affiche la révision
du micrologiciel, les heures de fonctionnement et lenombre de cycles de la batterie connectée.

surpendant que l’icône ANNULER est visible.

Mode Marche du mode avion confirmé

Page d’étatLes heures de fonctionnement sont indiquées en heuresentières.
Les décharges de la batterie correspondent au nombre
de cycles pendant lesquels la batterie connectée a été
vidangée et rechargée.

MODE AVIONLe FreeStyle Comfort est livré avec le mode avion
désactivé.

Pour activer le mode Arrêt du mode avion, répétez lesétapes ci-dessus tout en basculant sur ARRÊT.
Si vous appuyez surlorsque l’écran affiche lesymbole « Suivant », la page du menu passera àl’option d’écran suivante.
Le mode avion est destiné à être utilisé en moded’exécution. Le mode avion sera automatiquement
désactivé si l’appareil est laissé en veille lorsqu’il est
alimenté par la batterie.
Remarque : Consultez la section Activer le mode avion
pour en savoir plus sur le mode avion.

MODE DE SENSIBILITÉ

Écran principal du mode avion

Le FreeStyle Comfort est livré de l’usine avec un modede sensibilité par défaut défini sur 2 et ne nécessite
aucun réglage supplémentaire dans des conditions
normales. Les paramètres de sensibilité peuvent êtreréglés à des valeurs situées entre 1 et 5, 1 étant lavaleur moins sensible et 5 la plus sensible.

Pour activer le mode, appuyez sur
pendant quel’icône MODIFIER est visible. La valeur basculera surle mot bleu « MARCHE ».

Écran du mode de sensibilité défini sur 2

Mode Marche du mode avion sélectionné

Pour confirmer le mode Marche du mode avion,
appuyez sur
pendant que l’icône ENREGISTRER
est visible. La valeur basculera sur le mot noir
« MARCHE ».

60 - FR PN MN232-C4 L | Manuel d’utilisation

FreeStyle Comfort
Pour régler le mode de sensibilité, appuyez sur
pendant que l’icône MODIFIER est visible. La valeurbasculera sur le prochain numéro bleu.

Réglage du mode de sensibilité défini sur 3 sélectionné

Pour confirmer le réglage du mode de sensibilité
défini sur 3, appuyez sur
pendant que l’icône
ENREGISTRER est visible. La valeur basculera surle chiffre noir « 3 ».

MODE AUTOSATLe FreeStyle Comfort est livré avec le mode
autoSAT activé.

Écran principal du mode autoSAT

Pour désactiver le mode, appuyez sur
pendant quel’icône MODIFIER est visible. La valeur basculera surle mot bleu « ARRÊT ».

Pour annuler la modification du mode de sensibilité,
appuyez sur
est visible.

pendant que l’icône ANNULER

Mode Arrêt d’autoSAT sélectionné

Pour confirmer le mode Arrêt d’autoSAT, appuyez sur

Réglage du mode de sensibilité défini sur 3 confirmé

pendant que l’icône ENREGISTRER est visible.
La valeur basculera sur le mot noir « ARRÊT ».
Pour annuler le mode Marche d’autoSAT, appuyez sur
pendant que l’icône ANNULER est visible.

Réglage du mode de sensibilité défini sur 5 (max.) confirmé

Si vous appuyez surlorsque l’écran affichele symbole « Suivant », la page du menu passeraà l’option d’écran suivante.

Mode Arrêt d’autoSAT confirmé

Pour activer le mode Marche d’autoSAT, répétez lesopérations ci-dessus tout en basculant sur MARCHE.
Si vous appuyez surlorsque l’écran affichele symbole « Suivant », la page du menu passeraà l’option d’écran suivante.

PN MN232-C4 L | Manuel d’utilisation FR - 61

FreeStyle Comfort
Dépannage
Problème

Cause probable

Solution

L'appareil FreeStyle Comfort
affiche une alarme et émet un
bip intermittent.

Consultez le tableau relatif àl’état de l’alarme.

Consultez le tableau relatif à l’état de l’alarme.

Le concentrateur d’oxygène
ne s’allume pas.

La batterie est déchargée.

Alimentez l’appareil à l’aide d’une prise CA ou CC.

Non branché à une sourced’alimentation externe.

Assurez-vous que les branchements externes
soient sécurisés.

Dysfonctionnement général.

Effectuez un cycle à pleine puissance en retirantla batterie et en déconnectant toute sourced'alimentation externe CA ou CC. Ensuite,réinstallez la batterie et le cycle d'alimentation
est terminé. Reconnectez la source d'alimentation
externe si vous le souhaitez.
Contactez le fournisseur de votre équipement
et passez à une autre source d’alimentation en
oxygène si nécessaire.

Tous les autres problèmes.

Effectuez un cycle à pleine puissance en retirantla batterie et en déconnectant toute sourced'alimentation externe CA ou CC. Ensuite,réinstallez la batterie et le cycle d'alimentation
est terminé. Reconnectez la source d'alimentation
externe si vous le souhaitez.
Passez à une autre source d’oxygène et contactez
le fournisseur de votre équipement.

Remarque : Certaines alarmes entrainent l'arrêt de l'appareil.
Remarque : pendant une perte d'alimentation totale, le système d'alarme se réinitialise (efface toutes les alarmesprécédentes) et n'affiche une alarme que si la condition persiste une fois l'alimentation rétablie.

62 - FR PN MN232-C4 L | Manuel d’utilisation

FreeStyle Comfort
Informations relatives au transport
Le FreeStyle Comfort est conforme aux directives de la
FAA pour une utilisation à bord des avions commerciaux.
Avant tout voyage (avion, train, bateau, etc.) :• Veillez à ce que le concentrateur fonctionne
correctement.
• Contactez votre compagnie de transport au moins48 heures avant le départ afin de l’informer que vous
voyagerez avec un appareil FreeStyle Comfort.
Il est recommandé de toujours disposer de batteries
supplémentaires lors de vos déplacements. Pour unvoyage en avion, veuillez charger suffisamment labatterie, en tenant compte du temps passé au sol(vérification avant le vol, sécurité, etc.) et des escales.
La plupart des compagnies aériennes exigent quel’utilisateur dispose du nombre adéquat de batteries
pleinement chargées pour alimenter l’appareil durant
au moins 150 % de la durée maximum estimée duvoyage (temps de vol, temps au sol avant et après levol, correspondances et retards imprévus).

1. Allumez le système d’alimentation POC en utilisantla batterie UNIQUEMENT, lorsqu’aucune sourced’alimentation externe n’est connectée.

2. Activez le mode avion à l’aide de l’une desméthodes suivantes :a. Suivez les instructions pour activer le mode avion
dans la section des instructions de navigation dumenu.

Remarque : Si vous prenez l’avion, il est recommandé deprendre des batteries supplémentaires dans un sac detransport sécurisé.

RÉGLEMENTATION FAA

Les batteries de rechange doivent être protégées
individuellement pour éviter les courts-circuits en lesplaçant dans l'emballage d'origine ou en isolant lesbornes, par ex. en attachant du ruban adhésif sur lesbornes exposées ou en plaçant chaque batterie dans
un sac en plastique séparé ou dans une pochette deprotection, puis en les transportant uniquement dansles bagages à main.

b. Appuyez sur les boutons Haut et Bas etmaintenez-les enfoncés pendant 5 secondes jusqu’à
ce que vous entendiez un long bip et que le symbole
de l’avion apparaisse.

ACTIVER LE MODE AVION
Alimentation par la batterie
Les systèmes d'alimentation POC peuvent être utilisés
dans un avion suivant les procédures normales
d'utilisation en utilisant la batterie uniquement,
lorsqu'aucune source d'alimentation externe n'est
connectéePrise d'alimentation
Dans un avion, si une prise de courant est disponible,
opérationnelle et que son emplacement convient à une
utilisation comme source d'alimentation POC, elle peut
servir de batterie de secours aux batteries POC. Dansce cas, la procédure recommandée suivante concernant
la transition de la batterie vers l'alimentation électriquede l'avion doit être suivie :3. Lorsque le symbole de l’avion est affiché, branchez
l’alimentation POC.

Remarque : En raison des limitations aériennes,
les batteries ne se chargent pas en mode Avion.

PN MN232-C4 L | Manuel d’utilisation FR - 63

FreeStyle Comfort
Accessoires

INSERTION DE L’APPAREIL DANS LE SAC
DE TRANSPORT

Afin de garantir des performances et une sécurité
optimales, utilisez uniquement les accessoires
ci-dessous fournis par CAIRE par le biais de votre
fournisseur d’oxygène. L’utilisation de produits
différents pourrait nuire aux performances et/ou à lasécurité du concentrateur.

1. Insérez l’appareil au fond du sac, en alignant les
ouvertures sur les filtres et la batterie.

Accessoires standard pour l'appareil FreeStyle
ComfortSac de transport

MI396-1

Sac à dos

MI459-1

Bandoulière

MI397-1

Alimentation CA (tous les cordons
sont inclus)

PW036-1S

Cordon d’alimentation CC

CD041-1

Filtre pour grosses particules externes(Droite)FI226-1

Filtre pour grosses particules externes(Gauche)FI227-1

2. Fermez les fermetures éclair situées au bas de lapoche.

Accessoires fournis en option
Pack supplémentaire de batterie
8 cellules

BT034-1S

Batterie à 16 cellules

BT035-1S

Chargeur de bureau de la batterie

BT036-1

Sac de transport pour les accessoires

MI372-2

Remarque : si elles sont reprises ci-dessus, des options
supplémentaires peuvent être disponibles pour lescordons d’alimentation spécifiques aux pays. Contactez
CAIRE ou votre fournisseur d’oxygène si vous souhaitez
commander d’autres accessoires.

Instructions relatives au sac detransport
Pour insérer et sécuriser votre appareil FreeStyle
Comfort dans le sac de transport, respectez les
instructions suivantes.

64 - FR PN MN232-C4 L | Manuel d’utilisation

FIXATION DE LA BANDOULIÈRE
1. La bandoulière peut être attachée directement
à l'appareil ou au sac de transport.
2. Clipsez la bandoulière sur l'anneau de l'appareil
ou du sac.

FreeStyle Comfort
Élimination et réutilisation
Renvoyez toujours l'appareil FreeStyle Comfort,y compris tous les composants, à votre fournisseurd'équipement pour une mise au rebut appropriée.
Vous pouvez également contacter vos services
municipaux locaux pour obtenir des instructions
concernant l’élimination appropriée de la batterie.
L'appareil peut être réutilisé entre deux personnes,
mais doit être retourné au fournisseur de l'équipement
pour être retraité.

DEEE et RoHS
Ce symbole a pour objet d’inciter lepropriétaire de l’équipement à le retourner à un centre
de recyclage à la fin de son cycle de vie, conformément
à la directive relative aux déchets d’équipementsélectriques et électroniques (DEEE).

Matériaux en contact direct avec
l’utilisateur• Boîtier du concentrateur : PC/ABS• Panneau de commande du concentrateur : Polyester
EBG7 ou équivalent• Sortie d’oxygène, embout : Aluminium• Plaque signalétique : Lexan• Raccords du cordon : Polycarbonate/Chlorurede vinyle• Cordon d’alimentation : PVC, métal• Alimentation : Lexan 940 (polycarbonate)• Batterie : PC/ABS• Étiquettes sur l’alimentation et la batterie :Film polyester• Sac de transport du concentrateur : Cordura, Néoprène,
doublure en nylon• Bandoulière : Polypropylène, acétyle, vinyle, Cordura,
Néoprène

Nos produits sont conformes à la Directive sur lalimitation des substances dangereuses (RoHS). Ils ne
contiendront pas plus qu’une quantité infime de plombou d’autres substances dangereuses.
MISE EN GARDE : concernant l’élimination
adéquate, contactez le fournisseur oul’organisme local pour en savoir plus.
MISE EN GARDE : Toujours éliminer le tamis
moléculaire conformément à toutes lesréglementations fédérales, nationales et locales enmatière de contrôle environnemental.

PN MN232-C4 L | Manuel d’utilisation FR - 65

FreeStyle Comfort
Test CEM
Directives et déclaration du fabricant relatives aux émissions électromagnétiques
L’appareil FreeStyle Comfort est destiné à être utilisé dans l’environnement électromagnétique spécifié ci-dessous.
Le client ou l’utilisateur de l’appareil FreeStyle Comfort doit s’assurer que ces derniers sont utilisés dansun environnement de ce type.
Test d’émissions
Conformité
Environnement électromagnétique - Directives
Émissions RF
CISPR 11
Émissions RF
CISPR 11
Émissions de courant
harmonique CEI 61000-3-2
Fluctuations de tension/
papillotement CEI 61000-3-3Groupe 1

Classe B
Classe A
Conforme

66 - FR PN MN232-C4 L | Manuel d’utilisation

L'appareil FreeStyle Comfort utilise uniquement l’énergie RF pour son
fonctionnement interne. Par conséquent, ses émissions RF sont très
faibles et ne sont pas susceptibles de provoquer des interférences
avec des équipements électroniques situés à proximité.
Le FreeStyle Comfort est adapté à une utilisation dans tous lesétablissements, y compris les établissements domestiques et ceux
directement connectés au réseau public d'alimentation basse tension
qui alimentent les bâtiments utilisés à des fins domestiques.

FreeStyle Comfort
Directives et déclaration du fabricant relatives à l’immunité électromagnétique

L’appareil FreeStyle Comfort est destiné à être utilisé dans l’environnement électromagnétique spécifié ci-dessous.
Le client ou l’utilisateur de l’appareil FreeStyle Comfort doit s’assurer que ces derniers sont utilisés dans un environnement
de ce type.
Test d’immunité
Niveau de test
Niveau de conformité
Environnement électromagnétique Directives
CEI 60601
Environnement
Les sols doivent être en bois, en cimentélectromagnétique - Contact ± 8 kV
Contact ± 8 kV
ou carrelés. Si les sols sont revêtus d’unDirectives
matériau synthétique, le taux d’humidité
Décharge d'air ± 15 kV
Décharge d'air ± 15 kV
relative doit être d’au moins 30 %.
CEI 61000-4-2
Transitoiresélectriques rapides/
Salves
CEI 61000-4-4Surtension
CEI 61000-4-5

Creux de tension,
coupures brèves
et variations detension sur lescircuits d’alimentationélectrique
CEI 61000-4-11± 2 kV pour les lignes
d’alimentation± 1 kV pour les lignes
d’entrée/sortie± 1 kV entre fils de ligne± 2 kV pour les lignes
d’alimentation
S.O.

Mode commun ± 2 kV sur± 1 kV entre fils de ligne
les lignes AC
Différentiel ± 1 kV sur leslignes AC
Mode commun ± 2 kV
sur les lignes d'E/S
extérieures
UT de 0 % pour 0,5 cycle(0 °, 45 °, 90 °, 135 °,180 °, 225 °, 270 °,315 °)

UT de 0 % pour 0,5 cycle(0 °, 45 °, 90 °, 135 °,180 °, 225 °, 270 °,315 °)

UT de 0 % pour 1 cycle(0 °)UT de 0 % pour 1 cycle(0 °)UT de 70 % (baissede 30 % en UT) pendant
25/30 cycles (0 °)UT de 70 % (baissede 30 % en UT) pendant
25/30 cycles (0 °)UT de 0 % pour250/300 cycles (0 °)UT de 0 % pour250/300 cycles (0 °)La qualité de l'alimentation secteur doitêtre celle d'un établissement de soinsde santé professionnel typique et desenvironnements de soins à domicile.

La qualité de l'alimentation secteur doitêtre celle d'un établissement de soinsde santé professionnel typique et desenvironnements de soins à domicile.

La qualité de l'alimentation secteur doitêtre celle d'un établissement de soinsde santé professionnel typique et desenvironnements de soins à domicile.
Si l’utilisateur du FreeStyle Comfort
nécessite un fonctionnement continu
pendant les coupures d’alimentation, il est
recommandé que le FreeStyle Comfort
soit alimenté par une source d’alimentation
ininterruptible ou une batterie.

Les champs magnétiques de la fréquenced’alimentation doivent être à des niveaux
caractéristiques d’un emplacement typique
3A/mdans un établissement de soins de santéprofessionnel et dans les environnements
de soins à domicile.
REMARQUE : UT correspond à la tension de secteur CA avant l’application du niveau de test.

Champ magnétiquede la fréquence d’ali3A/m
mentation (50/60 Hz)
CEI 61000-4-8

REMARQUE : Dans certains cas, le FreeStyle Comfort peut être réinitialisé après une décharge électrostatique importante
et peut nécessiter un cycle à pleine puissance pour reprendre son fonctionnement normal. Pour en savoir plus,
reportez-vous à la section relative aux dépannages.

PN MN232-C4 L | Manuel d’utilisation FR - 67

FreeStyle Comfort
Directives et déclaration du fabricant relatives à l’immunité électromagnétique

L’appareil FreeStyle Comfort est destiné à être utilisé dans l’environnement électromagnétique spécifié ci-dessous. Le client
ou l’utilisateur de l’appareil FreeStyle Comfort doit s’assurer que ces derniers sont utilisés dans un environnement de ce
type.
Test d’immunité Niveau de test CEI 60601 Niveau de conformité
Environnement électromagnétique Directives
10 Vrms
RF conduites
10 Vrms
Les équipements de communication RF
portables et mobiles ne doivent pas êtreCEI 61000-4-6
150 kHz à 80 MHz
utilisés à proximité des composants del’appareil FreeStyle Comfort, y compris
les câbles, à une distance inférieure à
la distance de séparation recommandée
RF rayonnées
10 V/m
10 V/mcalculée sur la base de l’équation applicable
CEI 61000-4-3
80 MHz à 2700 MHz
à la fréquence de l’émetteur.
Distance de séparation recommandéed = 1.2 P

d = 1.2 P = 80 MHz à 800 MHz
d = 1.2 P = 800 MHz à 2 700 MHz
où P correspond à la puissance nominale desortie maximum de l’émetteur en watts (W),indiquée par le fabricant de l’émetteur etd correspond à la distance de séparation
recommandée en mètres (m).
L’intensité des champs émis par desémetteurs RF fixes, déterminée par l’étude
électromagnétique a doit être inférieur au
niveau de conformité pour chaque plage defréquences. b
Des interférences peuvent se produireà proximité des équipements portant lesymbole suivant :REMARQUE 1 : à 80 MHz et 800 MHz, la plage de fréquences supérieure s’applique.
REMARQUE 2 : il est possible que ces directives ne s’appliquent pas à toutes les situations. La propagationélectromagnétique dépend des phénomènes d’absorption et de réflexion induits par les structures, les objets
et les personnes.
aL’intensité des champs produits par des émetteurs fixes, tels que les stations de base pour les radiotéléphones (cellulaires/sans fil) et les radios mobiles, les radios amateurs, les radios AM et FM et la télévision ne peut pas être prédite
théoriquement avec précision. Pour évaluer l’environnement électromagnétique dû aux émetteurs RF fixes, une étude
électromagnétique du site doit être envisagée. Si l’intensité du champ, mesurée à l’emplacement où l'appareil FreeStyle
Comfort est utilisé, dépasse le niveau de conformité RF applicable indiqué ci-dessus, il conviendra de s’assurer du bon
fonctionnement de l'appareil FreeStyle Comfort dans un tel environnement. En cas d’anomalie, il pourra être nécessaire deprendre des mesures supplémentaires, par exemple en modifiant l’orientation ou la position de l’appareil FreeStyle Comfort.
b

Sur la plage de fréquences de 150 kHz à 80 MHz, l’intensité des champs doit être inférieure à 10 V/m.

68 - FR PN MN232-C4 L | Manuel d’utilisation

FreeStyle Comfort
Distances de séparation recommandées entre les équipements de communication
RF mobiles et portables et l’appareil FreeStyle Comfort

Le FreeStyle Comfort est prévu pour une utilisation dans un environnement dans lequel les perturbations dues aux
radiofréquences rayonnées sont contrôlées. Le client ou l’utilisateur de l’appareil FreeStyle Comfort peut contribuer
à prévenir les interférences électromagnétiques en maintenant une distance minimale entre les équipements decommunication RF portables (émetteurs) et le FreeStyle Comfort, comme indiqué ci-dessous, en fonction de la puissancede sortie maximale de l’équipement de communication.
Puissance nominale
Distance de séparation en fonction de la fréquence de l’émetteurde sortie maximummde l’émetteur
150 kHz à 80 MHz
80 MHz à 800 MHz
800 MHz à 2,5 GHz= 1.2 Pd = 1.2 P
d = 1.2 P
d
W
0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
Pour les émetteurs avec une puissance nominale de sortie maximum ne figurant pas dans le tableau ci-dessus, la distancede séparation recommandée (d) en mètres (m) peut être déterminée à l’aide de l’équation applicable à la fréquence
de l’émetteur, où P correspond à la puissance nominale de sortie maximum de l’émetteur en watts (W), indiquée par lefabricant de l’émetteur.
REMARQUE 1 : à 80 MHz et 800 MHz, la distance de séparation correspondant à la plage de fréquences supérieures’applique.
REMARQUE 2 : il est possible que ces directives ne s’appliquent pas à toutes les situations. La propagationélectromagnétique dépend des phénomènes d’absorption et de réflexion induits par les structures, les objets et lespersonnes.

Classification

Type de protection contre les chocs électriques :La protection de Classe II			contre les chocs électriques			est obtenue grâce à une double isolation.
Degré de protection contre les chocs électriques :Équipement de type BF fournissant un degré		de protection spécifique contre les chocs		électriques concernant		1) le courant de fuite autorisé ;		2) la fiabilité de la mise à la terre de protection		(le cas échéant).
		Non destiné aux utilisations cardiaques directes.
Tests indépendants pour la norme des équipementsélectromédicaux :	Testé pour sa conformité• IEC 60601-1 édition 3.1 : Équipement électriquemédical - Partie 1 Exigences générales pour lasécurité.
• Norme CAN/CSA C22.2 no 60601-1-08 édition
2.1 Équipement électrique médical – Partie 1 :exigences générales de sécurité
Protection contre les interférences électromagnétiques
et autres entre cet appareil et d’autres dispositifs.
• Testé pour être en conformité avec la norme
EN 60601-1-2 /CEI 60601-1-2 : 4e édition.

	• Testé pour être en conformité avec le document
RTCA/DO160
	Section 21 Catégorie M.
• CISPR 11 / EN 55011 Classe B Groupe 1,
	« Industriel, Scientifique et Médical (ISM)	médicaux » (ISM)• FCC Partie 15, Sous-partie B – Classe B
	Radiateurs involontaires
IP22 - Protection contre l'entrée de doigts ou d'objets
similaires d'un diamètre supérieur à 12,5 mm. L'eau
tombant verticalement ne doit pas avoir d'effet
néfaste lorsque l'habitacle est incliné à un angle de
15 degrés par rapport à sa position normale.
Mode autorisé de nettoyage et de prévention
des infections : veuillez consulter la section« Nettoyage, entretien et maintenance régulière »de ce manuel d’utilisation.
Degré de sécurité d’utilisation en présence de gazanesthésiques inflammables :	Appareil non adapté à ce type d’application.
Mode de fonctionnement : Fonctionnement en continu.

PN MN232-C4 L | Manuel d’utilisation FR - 69

FreeStyle Comfort
Steuerungselemente und Systemstatusanzeigen
ISO 7000
Bedienungsanleitung;
Betriebsanweisungen. Reg.-Nr. 1641
Land und Herstellungsdatum.
Das „CC“ gibt den zweibuchstabigen
Ländercode des Herstellungslandes
an. Das Herstellungsdatum ist im
Format JJJJ-MM-TT. Reg.-Nr. 6049

ISO 7010: Graphische Symbole –Sicherheitsfarben und Sicherheitszeichen –Registrierte Sicherheitszeichen
Die Betriebsanleitung ist aufmerksam
durchzulesen. Reg.-Nr. M002
Von Flammen, offenem Feuer
und Funken fernhalten. Offene
Zündquellen und Rauchen verboten.
Reg.-Nr. P003
Nicht in der Nähe des Geräts oderwährend dessen Betrieb rauchen.
Reg.-Nr. P002

Vor Regen schützen, trocken
aufbewahren. Reg.-Nr. 0626
Stapelgrenze nach Anzahl, wobei„n“ die maximale Anzahl zulässiger
Einheiten angibt. Reg.-Nr. 2403
Katalognummer. Reg.-Nr. 2493

Warnung. Reg.-Nr. W001
IEC 60417
Anwendungsteil des Typs BF(Schutzgrad gegen Stromschlag).
Reg.-Nr. 5333

Seriennummer. Reg.-Nr. 2498
Lager- oder Betriebstemperaturbereich.
Reg.-Nr. 0632
Luftfeuchtigkeitsbereich für die
Lagerung. Reg.-Nr. 2620
Luftdruckbegrenzung. Reg.-Nr. 2621
Name und Anschrift des Herstellers.
Reg.-Nr. 3082
Diese Seite nach oben. Reg.-Nr. 0623

Gerät der Klasse II, doppelt isoliert
Reg.-Nr. 5172
EU-Verordnung 2017/745
Autorisierter Vertreter in derEuropäischen Gemeinschaft
Wenn das UDI-Etikett (Unique Device
Identifier) das Symbol CE####
enthält, erfüllt das Produkt die
Anforderungen an Medizinprodukte.
Das CE####-Symbol zeigt die
Nummer der benannten Stelle an.
Zusätzliche Symbole
Von entzündlichen Materialien,Öl und Fett fernhalten.

Zerbrechlich, Vorsicht bei derHandhabung. Reg.-Nr. 0621

Nicht zerlegen.

Enthält gefährliche Substanzen.
Reg.-Nr. 3723

Durchfluss erhöhen.

Importeur. Reg.-Nr. 3725
Durchfluss reduzieren.
Sauerstoffausgang

Menü-Taste.
Netzschalter. Reg.-Nr. 5009
Konform
mit RTCA/
DO-160Abschnitt 21
Kategorie M

70 - DE PN MN232-C4 L | Bedienungsanleitung

Der Hersteller dieses POC hat
festgestellt, dass dieses Gerät allen
geltenden FAA-Akzeptanzkriterien
und der Verwendung an Bord vonFlugzeugen entspricht.

FreeStyle Comfort
Der Hersteller dieses POC hat
festgestellt, dass dieses Gerät allen
geltenden FAA-Akzeptanzkriterien
und der Verwendung an Bord vonFlugzeugen entspricht.
Recycling-Symbol
Bevollmächtigter für die Schweiz(ADD)
Wenn das Gerät das UKCA-Zeichen
trägt, wobei UKCA#### die
Nummer der benannten Stelleanzeigt, entspricht dieses Gerät den
Vorschriften der UKCA.
21 CFR 801.15
Nach den in den USA geltenden
Gesetzen darf dieses Gerät nur voneinem Arzt oder im Auftrag eines
Arztes verkauft werden.

Richtlinie 2012/19/EU des Europäischen
Parlaments und des Rates
WEEE-Richtlinie
Das Symbol weist den Besitzer
des Geräts darauf hin, dass das
Gerät gemäß der Richtlinie überElektro- und Elektronik-Altgeräte
am Ende seiner Lebensdauer zu einer
Recyclingstelle gebracht werdenmuss. Unsere Produkte entsprechen
der Richtlinie zur Beschränkung der
Verwendung bestimmter gefährlicher
Stoffe (RoHS). Sie enthalten Blei
oder andere gefährliche Materialien
höchstens in Spuren.
IEC 60601-1-8
Technischer Alarm mit niedriger
Priorität
ISO 15223-1

ETL-Zertifizierung
Zertifiziert für den US-amerikanischen
und den kanadischen Markt, gemäß
den geltenden US-amerikanischen und
kanadischen Normen
Enthält die FCC-ID: WAP2001
FCC-Hinweis
Dieses Gerät kann in Übereinstimmung mit Par.
15 der FCC-Regeln CYBLE-022001-00 enthalten,
einschließlich der Antenne 2450AT18B100
von Johnson Technology. Das Gerät erfüllt
die Anforderungen für zugelassene modulare
Sender, wie in der FCC-Mitteilung DA00-1407
beschrieben. Der Betrieb des Senders unterliegt
den folgenden zwei Bedingungen: (1) Das Gerät
darf keine schädlichen Interferenzen verursachen,und (2) Dieses Gerät muss alle empfangenen
Interferenzen akzeptieren, einschließlich
Interferenzen, die einen unerwünschten Betrieb
verursachen können. Das Produkt kann durch
ein oder mehrere Patente geschützt sein, sowohl
in den USA als auch international. Auf unserer
Webseite finden Sie eine Auflistung der geltenden
Patente. Pat: www.caireinc.com/corporate/patents

Medizinprodukt. Reg.-Nr. 5.7.7
Eindeutige Gerätekennung
Reg.-Nr. 5.7.10
A S/NZS 4417.1, AS/NZS 4417.2
ACMA Regulatory ComplianceMark – zeigt die Einhaltung der
ACMA-Vorschriften (ADD) an
Funktechnologie
Verbindungsarten

SSP, iAP2, GATT

Frequenz

2.402 bis 2.480 MHz

Max. HF-Ausgangsleistung+4 dBmBetriebsbereich

10 m (Klasse 2)

Während des Betriebs sollte sich das Gerät ineiner Entfernung von mindestens 1,1 cm (0,43 Zoll)
zum Körper befinden. Dies gilt nicht für Masken,
Luftschläuche und Zubehör.

Grafische Symbole zur Verwendung auf Geräten
nach IEC 60417
Gleichstrom Reg.-Nr. 5031
Wechselstrom Reg.-Nr. 5032
IEC 60601-1
Tropfwassergeschützt

PN MN232-C4 L | Bedienungsanleitung DE- 71

FreeStyle Comfort
Auspacken Ihres FreeStyle ComfortÜberprüfen Sie, ob alleaufgeführten Komponenten
im Paket enthalten sind.
Falls etwas fehlen sollte,
setzen Sie sich sofort mit
Ihrem Sauerstoffanbieter in
Verbindung.

Standardzubehör• Tragbarer Sauerstoffkonzentrator• Tragetasche• Schultergurt• 8-Zellen Akkupack• Wechselstrom-Netzteil• Wechselstromkabel• Gleichstrom-Netzteil• Benutzerhandbuch

Optionales Zubehör• Zusätzlicher 8-Zellen Akkupack• 16-Zellen-Akkupack• Tischladegerät• Zubehörtragetasche

Einführende Informationen zu Ihrem FreeStyle Comfort
Machen Sie sich mit den wichtigsten Funktionen des FreeStyle Comfort und der
Benutzeroberfläche vertraut.
Lasche für den
Schultergurt

Vorderansicht

Benutzeroberfläche

Sauerstoffauslass
Luftauslass

Lufteinlass/Filter
Lufteinlass/Filter

Eingang fürden Stromanschluss
Akkupack
Akkupack-Entriegelungsknopf
72 - DE PN MN232-C4 L | Bedienungsanleitung

FreeStyle Comfort
Draufsicht
Bildschirm
Durchfluss
erhöhenEin-/Aus-Schalter

Menü-Taste

Durchfluss
reduzieren

Bildschirm bei Akkubetrieb
Einstellung derDurchflussrate

Akkuanzeige

autoSAT-Anzeiger
Alarmanzeige

Akkustandin Prozent

Anzeige
Bolusgabe

Bildschirm bei Netzbetrieb
Einstellung derDurchflussrate

Alarmanzeigefeld

Akkuanzeige

Ladeanzeige

PN MN232-C4 L | Bedienungsanleitung DE- 73

FreeStyle Comfort
Was genau ist der
Sauerstoffkonzentrator?
Die Atemluft enthält ca. 21 % Sauerstoff, 78 %Stickstoff und 1 % andere Gase. Beim FreeStyle®
Comfort® wird die Raumluft durch die Lufteinlässe
in das Gerät gesaugt. Sie fließt dann durch ein
Absorbermaterial, das sogenannte Molekularsieb.
Dieses Material trennt den Sauerstoff vom
Stickstoff und lässt nur den Sauerstoff hindurch.
Das Ergebnis ist hochreiner Sauerstoff, der dem
Anwender zugeführt wird.
Im FreeStyle Comfort werden
fortschrittliche Sauerstoffkonzentrator- undSauerstoffkonservierungs-Techniken kombiniert.
Das tragbare Gerät bietet maximale Portabilität
und Mobilität. Das fortschrittliche ImpulsflussAbgabesystem bemerkt schnell, wenn der
Benutzer einatmet, und gibt jeweils zu Beginn
der Einatemphase einen Sauerstoffimpuls ab. Das
Impulsfluss-Abgabesystem ist ideal für Aktivitäten
außerhalb des eigenen Hauses. Er konserviert
den Sauerstoff und verlängert im Gegensatz zum
kontinuierlichen Durchfluss die Lebensdauer
des Akkus. Der FreeStyle Comfort ist leicht undbequem per Schultergurt zu tragen. Es kann mit
Wechselstrom (Wandsteckdose), Gleichstrom(Fahrzeug) oder mit dem Akku betrieben werden.

Bestimmungsgemäße Verwendung
Der CAIRE FreeStyle ComfortSauerstoffkonzentrator ist für die Verabreichungvon zusätzlichem Sauerstoff bestimmt. Das
Gerät ist nicht für lebenserhaltende Maßnahmen
ausgelegt und bietet keine Patientenüberwachung.

Gründe für die Verschreibung vonSauerstoff durch den Arzt
Viele Menschen leiden unter
verschiedenartigen Herz-, Lungen- und anderen
Atemwegserkrankungen. Einem bedeutenden
Teil dieser Menschen kann durch eine zusätzliche
Sauerstoffzufuhr zu Hause, auf Reisen oder bei
Tagesaktivitäten geholfen werden.
Sauerstoff ist ein Gas, das 21 % der Umgebungsluft
ausmacht, die wir atmen. Unser Körper ist auf
eine konstante Zufuhr angewiesen, um richtig zu
funktionieren. Ihr Arzt hat Ihnen eine Flussrateoder eine Einstellung gemäß Ihrem persönlichen
respiratorischen Zustand verschrieben.

74 - DE PN MN232-C4 L | Bedienungsanleitung

Obgleich Sauerstoff als Medikament nicht
süchtig macht, kann eine nicht verordnete
Sauerstofftherapie gefährlich sein. Vor der
Verwendung dieses Sauerstoffkonzentrators müssen
Sie einen Arzt konsultieren. Der Geräteanbieter
Ihres Sauerstoffgeräts wird Ihnen zeigen, wie Sie
die verschriebene Flussrate einstellen.

Profil des Anwenders
Konzentratoren sind für die zusätzliche
Sauerstoffzufuhr an Anwender bestimmt,
die unter Beschwerden aufgrund eines
beeinträchtigten Übergangs des Sauerstoffs vonder Lunge in den Blutkreislauf leiden. Tragbare
Sauerstoffkonzentratoren (POCs) speichern undenthalten keinen Sauerstoff. Sie müssen nicht
wieder aufgefüllt werden und können überall
aufgeladen werden, wo Wechsel- oder Gleichstromvorhanden ist (außer im Flugzeug). Auf diese Weise
ist der Benutzer relativ unabhängig in Bezug auf
Flexibilität (sowohl zu Hause als auch unterwegs),
Mobilität und in seinem gesamten Lebensstil.
Die Verwendung eines Sauerstoffkonzentrators
erfordert eine ärztliche Verordnung.
Der typische Anwender einer Sauerstofftherapie
ist älter als 65 Jahre und leidet an einer Reihe
von Atemwegserkrankungen, einschließlich
der chronisch obstruktiven Lungenerkrankung(COPD). Üblicherweise verfügen die Anwendernoch über gute kognitive Fähigkeiten und müssen
dazu in der Lage sein, Unwohlsein mitzuteilen.
Ist der Anwender nicht (mehr) in der Lage,
Unwohlsein mitzuteilen oder die Aufkleber auf
dem Konzentrator bzw. die Gebrauchsanweisungzu lesen und zu verstehen, darf das Gerät nur
unter Aufsicht einer Person verwendet werden,
die hierzu in der Lage ist. Treten während derVerwendung eines Konzentrators Beschwerden auf,sollten sich die betroffenen Anwender mit ihrem
Arzt in Verbindung setzen. Es wird Anwendern
empfohlen, Reservesauerstoff (z. B. in Form einer
Sauerstoffflasche) bereitzuhalten, der im Falleeines Stromausfalls oder eines sonstigen Ausfalls
des Konzentrators verwendet werden kann.
Weitere besondere Fertigkeiten oder Fähigkeiten
des Anwenders sind für den Gebrauch desSauerstoffkonzentrators nicht erforderlich.

FreeStyle Comfort
Wichtig!
Warn- und Sicherheitshinweise sind folgendermaßen
gekennzeichnet:
WARNUNG: WEIST AUF GEFAHREN
HIN, DIE SCHWERE VERLETZUNGENZUR FOLGE HABEN KÖNNEN.
VORSICHT: Wichtige Informationen,um Schäden am FreeStyle Comfort
zu vermeiden.
Hinweis: Besonders zu beachtende Informationen.

Gebrauchshinweise
WARNUNG: DAS VERORDNETESAUERSTOFFNIVEAU MUSS
UNBEDINGT EINGEHALTEN WERDEN.
EINSTELLUNGEN DER FLUSSRATEDÜRFEN NUR GEÄNDERT WERDEN,
WENN DIES AUF ANWEISUNG EINESAPPROBIERTEN ARZTES ERFOLGT.
WARNUNG: DIESES PRODUKT WURDE NICHT ZUR
VERWENDUNG MIT PÄDIATRISCHEN PATIENTEN
GETESTET ODER ZUGELASSEN.
WARNUNG: LAUT US-GESETZEN DARFDIESES GERÄT NUR AUF ANWEISUNG EINESARZTES ODER EINES ANDEREN LIZENZIERTENGESUNDHEITSDIENSTLEISTERS VERKAUFTODER VERLIEHEN WERDEN.
WARNUNG: DIESES GERÄT DARF NICHT FÜR
LEBENSERHALTENDE MASSNAHMEN EINGESETZT WERDEN. FÜR BENUTZER, DIE BEI DER
VERWENDUNG DIESES GERÄTS NICHT IN DER
LAGE SIND, BESCHWERDEN ZU ÄUSSERN, KANN
EINE ZUSÄTZLICHE ÜBERWACHUNG ERFORDERLICH SEIN. ANWENDER MIT EINGESCHRÄNKTEM
HÖR- UND/ODER SEHVERMÖGEN BENÖTIGEN
MÖGLICHERWEISE HILFE BEIM ÜBERWACHENDER ALARME.
WARNUNG: DIE EINSTELLUNGEN FÜR DEN IMPULSDOSIS-MODUS MÜSSEN FÜR JEDEN NUTZERINDIVIDUELL ENTSPRECHEND DEM PERSÖNLICHEN BEDARF IN RUHEPOSITION, BEI BEWEGUNGODER AUF REISEN ANGEPASST WERDEN.
WARNUNG: DIESES GERÄT IST FÜR DIE VERWENDUNG IN DER HÄUSLICHEN PFLEGE UND ÄHNLICHEN AMBULANTEN UMGEBUNGEN VORGESEHEN.

WARNUNG: EINIGE ATEMANSTRENGUNGEN DESPATIENTEN LÖSEN MÖGLICHERWEISE NICHT DIE
KONSERVIERUNGSFUNKTION AUS.
WARNUNG: DER FREESTYLE COMFORT IST
NICHT FÜR DIE VERWENDUNG MIT EINEM
TRACHEOTOMIERTEN PATIENTEN VORGESEHEN.
WARNUNG: SCHWANGERE ODER STILLENDEFRAUEN SOLLTEN KEINE IN DIESEM HANDBUCH
EMPFOHLENEN ZUBEHÖRTEILE VERWENDEN, DA
DIESE PHTHALATE ENTHALTEN KÖNNEN.

Gegenanzeigen
WARNUNG: DIE VERWENDUNG VONSAUERSTOFF OHNE ÄRZTLICHEVERSCHREIBUNG IST UNTER
UMSTÄNDEN GEFÄHRLICH. DAS GERÄTDARF NUR AUF VERSCHREIBUNGEINES ARZTES VERWENDET WERDEN.
WARNUNG: DIESES PRODUKT IST NICHT FÜR
DIE VERWENDUNG IN ODER IN DER NÄHE VON
MAGNETRESONANZUMGEBUNGEN, WIE
MRT-GERÄTEN, VORGESEHEN.
WARNUNG: DAS GERÄT DARF NICHTVERWENDET WERDEN, WENN BRENNBARE
ANÄSTHETIKA VORHANDEN SIND.
WARNUNG: WIE ALLE ELEKTRISCH
BETRIEBENEN GERÄTE KANN AUCH DERSAUERSTOFFKONZENTRATOR, BEISPIELSWEISE
BEI EINEM STROMAUSFALL ODER WENN EINEWARTUNG DURCH EINEN QUALIFIZIERTENTECHNIKER ERFORDERLICH IST, AUSFALLEN.
DER SAUERSTOFFKONZENTRATOR IST NICHTFÜR ANWENDER GEEIGNET, DIE DURCH EINESOLCHE TEMPORÄRE UNTERBRECHUNGGESUNDHEITLICHEN SCHADEN NEHMEN WÜRDEN.
WARNUNG: DIESES PRODUKT IST NICHT FÜR
DIE VERWENDUNG IN ALLEN UMGEBUNGENZUGELASSEN. DIE EXPOSITION GEGENÜBERBESTIMMTEN ELEKTROMAGNETISCHENQUELLEN KANN ZU EINER INTERFERENZ ODERUNBEABSICHTIGTEN LEISTUNG DES PRODUKTSFÜHREN. BEI EINER STÖRUNG MUSS DAS GERÄT
SOFORT AUSGESCHALTET WERDEN.

WARNUNG: DIESES GERÄT IST FÜR DIE
VERWENDUNG DURCH DEN PATIENTEN BESTIMMT,DER EINE SAUERSTOFFTHERAPIE ERHÄLT.

PN MN232-C4 L | Bedienungsanleitung DE- 75

FreeStyle Comfort
Sicherheitsrichtlinien
WARNUNG: AN DIESEM GERÄT
DÜRFEN KEINE MODIFIKATIONEN
VORGENOMMEN WERDEN.
WARNUNG: DIESES GERÄT LIEFERT HOCHKONZENTRIERTEN SAUERSTOFF, DER SCHNELLZU BRÄNDEN FÜHREN KANN. RAUCHEN ODER
OFFENE FLAMMEN SIND IM RAUM, IN DEM SICH (1)DAS GERÄT ODER (2) EIN SAUERSTOFF TRANSPORTIERENDES ZUBEHÖRTEIL BEFINDEN, NICHTERLAUBT. DIE MISSACHTUNG DIESER WARNUNG
KANN SCHWERE BRÄNDE, SACH- BZW. PERSONENSCHÄDEN ODER DEN TOD ZUR FOLGE HABEN.
WARNUNG: BETREIBEN SIE DAS GERÄT NICHTIN EINEM EINGESCHRÄNKTEN ODER ENGEN BEREICH (Z. B. EINEM KLEINEN KOFFER ODER EINERHANDTASCHE), IN DEM DIE BELÜFTUNG UNZUREICHEND SEIN KÖNNTE. HIERDURCH KANN SICH
DER SAUERSTOFFKONZENTRATOR ÜBERHITZENUND DIE LEISTUNG BEEINTRÄCHTIGT WERDEN.
WARNUNG: DER KONZENTRATOR DARF NICHTMIT RAUCH, SCHADSTOFFEN ODER DÄMPFEN
IN KONTAKT KOMMEN.
WARNUNG: FALLS EIN SCHWERWIEGENDERVORFALL MIT DIESEM GERÄT AUFTRITT, SOLLTE
DER BENUTZER DEN VORFALL SOFORT DEM ANBIETER UND/ODER DEM HERSTELLER MELDEN.
EIN SCHWERWIEGENDER VORFALL IST DEFINIERT ALS VERLETZUNG, TOD ODER POTENZIELLE VERLETZUNGS-/TODESURSACHE FÜR DEN
FALL, DASS SICH DER VORFALL WIEDERHOLT.
DER BENUTZER KANN DEN VORFALL AUCH DERZUSTÄNDIGEN BEHÖRDE DES LANDES MELDEN,
IN DEM SICH DER VORFALL EREIGNET HAT.
WARNUNG: DER EINSATZ EINIGER SAUERSTOFF FÜHRENDER ZUBEHÖRTEILE,DIE NICHT FÜR DIE VERWENDUNG MIT
DIESEM SAUERSTOFFKONZENTRATORVORGESEHEN SIND, KANN DESSEN
LEISTUNG BEEINTRÄCHTIGEN. AUF
EMPFOHLENES ZUBEHÖR WIRD IN
DIESER ANLEITUNG VERWIESEN.
WARNUNG: WENN DER SAUERSTOFFKONZENTRATOR HERUNTERGEFALLEN IST, BESCHÄDIGTWURDE ODER MIT WASSER IN BERÜHRUNG GEKOMMEN IST, WENDEN SIE SICH BITTE AN IHRENAUSRÜSTER, UM DAS GERÄT ÜBERPRÜFEN UND
GEGEBENENFALLS REPARIEREN ZU LASSEN.
VERWENDEN SIE DEN SAUERSTOFFKONZENTRATOR NICHT, WENN STROMKABEL ODER STECKER
BESCHÄDIGT SIND.
WARNUNG: SICHERN SIE DEN SAUERSTOFFKONZENTRATOR MIT EINEM GURT ODER ÄHNLICHENVORRICHTUNGEN, WENN ER IN EINEM FAHRZEUG TRANSPORTIERT WIRD, UM SCHÄDEN
BZW. VERLETZUNGEN ZU VERMEIDEN.
76 - DE PN MN232-C4 L | Bedienungsanleitung

WARNUNG: DIE RICHTLINIEN DES VERKEHRSMINISTERIUMS DER USA (DOT) SOWIE DER
VEREINTEN NATIONEN (UN) SEHEN VOR, DASSDER AKKU AUS DEM GERÄT HERAUSGENOMMEN
WERDEN MUSS, WENN DIESES ALS GEPÄCK AUF
INTERNATIONALEN FLÜGEN AUFGEGEBEN WIRD.
BEI VERSAND DES SAUERSTOFFKONZENTRATORS MUSS DER AKKU EBENFALLS AUS DEM
GERÄT ENTNOMMEN UND ORDNUNGSGEMÄSS
VERPACKT WERDEN.
WARNUNG: DAS GERÄT NUR NACH
EINGEHENDEM STUDIUM UND VERSTÄNDNISDIESER ANLEITUNG VERWENDEN. SOLLTENSIE DIE WARNUNGEN UND ANWEISUNGENNICHT VERSTEHEN, WENDEN SIE SICH AN
DEN BEREITSTELLER DIESES GERÄTES,BEVOR SIE VERSUCHEN, ES ZU VERWENDEN;
ANDERNFALLS KANN ES ZU VERLETZUNGENODER SCHÄDEN KOMMEN.
WARNUNG: BEI UNWOHLSEIN ODER EINEM
MEDIZINISCHEN NOTFALL SOFORT ÄRZTLICHE
HILFE ZURATE ZIEHEN.
WARNUNG: WIRD DER SAUERSTOFFKONZENTRATOR AUSSERHALB DES ANGEGEBENEN TEMPERATURBEREICHES VERWENDET, KANN DIESDIE FÄHIGKEIT DES GERÄTES BEEINTRÄCHTIGEN, DIE ANGEGEBENEN SAUERSTOFFKONZENTRATIONEN ZU ERZEUGEN. FÜR TEMPERATURGRENZWERTE SIEHE ABSCHNITT „TECHNISCHEDATEN“ IN DIESER ANLEITUNG.
WARNUNG: KEINE ÖL-, FETT-, ODER PETROLEUMBASIERTEN ODER ANDEREN ENTZÜNDLICHEN PRODUKTE MIT DEM SAUERSTOFFKONZENTRATOR ODER DESSEN SAUERSTOFFBEFÖRDERNDEM ZUBEHÖR VERWENDEN. ES
DÜRFEN AUSSCHLIESSLICH WASSERBASIERTE,SAUERSTOFFVERTRÄGLICHE LOTIONEN ODERSALBEN VERWENDET WERDEN. SAUERSTOFFBESCHLEUNIGT DIE VERBRENNUNG ENTZÜNDLICHER SUBSTANZEN.
WARNUNG: EIN NICHT ORDNUNGSGEMÄSSERGEBRAUCH DES AKKUS KANN ZU EINEMERHITZEN UND ENTZÜNDEN DES AKKUSFÜHREN, WAS SCHWERE VERLETZUNGEN ZUR
FOLGE HABEN KANN. DURCHBOHREN SIE DENAKKU NICHT UND SCHLAGEN UND TRETEN SIE
NICHT AUF DEN AKKU. LASSEN SIE DEN AKKUNICHT FALLEN UND SETZEN SIE DEN AKKUAUCH KEINEN SONSTIGEN HEFTIGEN SCHLÄGENODER STÖSSEN AUS. DIE VERWENDUNGEINES BESCHÄDIGTEN AKKUS KANN ZU
VERLETZUNGEN FÜHREN.
WARNUNG: DIE METALLKONTAKTE DES
AKKUS DÜRFEN NICHT MIT METALLISCHENGEGENSTÄNDEN, Z. B. SCHLÜSSELN ODERMÜNZEN, KURZGESCHLOSSEN WERDEN. DAS
KANN ZU FUNKEN ODER ÜBERHITZUNG FÜHREN.

FreeStyle Comfort
WARNUNG: WIRD DER TRAGBARE SAUERSTOFFKONZENTRATOR IM FREIEN ÜBER WECHSELSTROM BETRIEBEN, SCHLIESSEN SIE DASSTROMKABEL NUR AN EINE STROMQUELLE MIT
FEHLERSTROM-SCHUTZEINRICHTUNG (RCD) AN.
WARNUNG: DER SAUERSTOFFKONZENTRATORSOLLTE NICHT DIREKT NEBEN ANDERENGERÄTEN ODER MIT ANDEREN GERÄTENGESTAPELT VERWENDET WERDEN. IST EINESOLCHE SITUATION NICHT ZU VERMEIDEN,MUSS DAS GERÄT AUF ORDNUNGSGEMÄSSEN
BETRIEB HIN ÜBERWACHT WERDEN.
WARNUNG: DIE VERWENDUNG VONANDEREN KABELN UND ADAPTERN ALS
DEN ANGEGEBENEN, MIT AUSNAHME DER
KABEL UND ADAPTER, DIE VOM HERSTELLER
DES MEDIZINISCHEN ELEKTRISCHEN
GERÄTS ALS ERSATZTEILE FÜR INTERNE
KOMPONENTEN VERKAUFT WERDEN,
KANN ZU ERHÖHTEN EMISSIONEN ODEREINER REDUZIERTEN STÖRFESTIGKEIT DESSAUERSTOFFKONZENTRATORS FÜHREN.
WARNUNG: VERWENDEN SIE NUR DIE
ELEKTRISCHE SPANNUNG, DIE AUF DEM
GERÄTEETIKETT MIT DEN TECHNISCHEN DATENANGEGEBEN IST.
WARNUNG: KEINE VERLÄNGERUNGSKABELFÜR DIESES GERÄT VERWENDEN UND NICHT ZU
VIELE STECKER MIT DERSELBEN STECKDOSEVERBINDEN. DIE VERWENDUNG EINES VERLÄNGERUNGSKABELS KANN DIE LEISTUNG DESGERÄTES BEEINTRÄCHTIGEN. DIE VERBINDUNGZU VIELER STECKER MIT EINER STECKDOSE
KANN DEN SCHALTSCHRANK ÜBERLASTEN UNDZUM AUSLÖSEN VON TRENNSCHALTER/SICHERUNG ODER BEI FEHLFUNKTION VON TRENNSCHALTER/SICHERUNG ZUM AUSBRUCH EINESFEUERS FÜHREN.
VORSICHT: Falls ein Alarm auftritt oder
Sie feststellen, dass der Sauerstoffkonzentrator nicht ordnungsgemäß
funktioniert, lesen Sie den Abschnitt
Fehlerbehebung in diesem Handbuch.
An den Geräteanbieter wenden, wenndas Problem nicht gelöst werden kann.

WARNUNG: NICHT VERSUCHEN, DASGERÄT – ABGESEHEN VON DEN IN
DIESER ANLEITUNG AUFGEFÜHRTENSTÖRUNGSBEHEBUNGEN – ZU WARTEN.
WARNUNG: VERWENDEN SIE NUR VOM
HERSTELLER EMPFOHLENES ZUBEHÖR.
DIE VERWENDUNG ANDEREN ZUBEHÖRS KANNZU EINER SCHWEREN BESCHÄDIGUNG DESSAUERSTOFFKONZENTRATORS SOWIE ZUM
ERLÖSCHEN DER GARANTIE FÜHREN.
WARNUNG: FÜR DEN FALL EINESSTROMAUSFALLS, ALARMS ODERMECHANISCHEN VERSAGENS EMPFIEHLTDER HERSTELLER EINE ALTERNATIVESAUERSTOFFZUFUHR. DEN ARZT ODERGERÄTEANBIETER FRAGEN, WELCHE ART
VON RESERVESYSTEM BENÖTIGT WIRD.
WARNUNG: STELLEN SIE SICHER, DASSLUFTEINLASS UND -AUSLASS NICHT BLOCKIERTWERDEN. FÜHREN SIE KEINE GEGENSTÄNDEIN DIE GERÄTEÖFFNUNGEN EIN UND LASSENSIE NICHTS IN DIESE HINEINFALLEN. HIERDURCHKANN SICH DER SAUERSTOFFKONZENTRATORÜBERHITZEN UND DIE LEISTUNG
BEEINTRÄCHTIGT WERDEN.
WARNUNG: DAS RAUCHEN WÄHREND DER
SAUERSTOFFVERABREICHUNG IST DER
HAUPTGRUND FÜR BRANDVERLETZUNGEN
UND -TODE. DIESE SICHERHEITSWARNUNGENMÜSSEN UNBEDINGT BEFOLGT WERDEN:
WARNUNG: RAUCHEN, KERZEN ODER OFFENE
FLAMMEN SIND IN DEM RAUM, IN DEM SICH DASGERÄT ODER EIN SAUERSTOFF BEFÖRDERNDESZUBEHÖRTEIL BEFINDEN, VERBOTEN.
WARNUNG: RAUCHEN WÄHREND DES TRAGENSEINER SAUERSTOFFKANÜLE KANN ZUGESICHTSVERBRENNUNGEN UND ZUM TODEFÜHREN.
WARNUNG: WENN SIE RAUCHEN, KÖNNEN
DIESE 3 SCHRITTE IHR LEBEN RETTEN: DEN
SAUERSTOFFKONZENTRATOR AUSSCHALTEN,
DIE KANÜLE ABNEHMEN UND DEN RAUM, IN DEM
DAS GERÄT SICH BEFINDET, VERLASSEN.

VORSICHT: Alle Anweisungen des Herstellers
befolgen, damit die Garantie ihre Gültigkeit behält.

PN MN232-C4 L | Bedienungsanleitung DE- 77

FreeStyle Comfort
WARNUNG: SCHILDER MIT DER AUFSCHRIFT„RAUCHEN VERBOTEN – SAUERSTOFF IN GEBRAUCH“ MÜSSEN IM HAUS ODER AN DEM ORT,
AN DEM DER SAUERSTOFF VERWENDET WIRD,
DEUTLICH ERKENNBAR ANGEBRACHT SEIN.
NUTZER UND PFLEGEPERSONAL MÜSSEN ÜBERDIE GEFAHREN DES RAUCHENS IN DER NÄHEODER WÄHREND DES GEBRAUCHS VON MEDIZINISCHEM SAUERSTOFF AUFGEKLÄRT WERDEN.
WARNUNG: DEN SAUERSTOFFKONZENTRATORNICHT BEI VORHANDENSEIN ENTZÜNDLICHERGASE VERWENDEN. DIES KANN SCHNELLZU BRÄNDEN FÜHREN, DIE SACHSCHÄDEN,VERLETZUNGEN ODER DEN TOD NACH SICHZIEHEN KÖNNEN.
WARNUNG: EIN ABNEHMEN DER KANÜLE UND
EINE LAGERUNG AUF KLEIDUNG, BETTWÄSCHE,SOFAS ODER ANDEREM POLSTERMATERIALFÜHREN BEI VORHANDENSEIN EINER ZIGARETTE,WÄRMEQUELLE ODER FLAMME ZU EINERVERPUFFUNG.
WARNUNG: LASSEN SIE EINE KANÜLE NICHT AUF
ODER UNTER KLEIDUNG, BETTWÄSCHE ODERSTUHLPOLSTERN LIEGEN. WENN DAS GERÄT
EINGESCHALTET IST, ABER NICHT VERWENDETWIRD, KANN DER SAUERSTOFF ENTZÜNDLICHWIRKEN. STELLEN SIE SICHER, DASS DER
SAUERSTOFFKONZENTRATOR AUSGESCHALTET
IST, WENN ER NICHT VERWENDET WIRD.
WARNUNG: ZUGELASSENES ZUBEHÖR ZUR
SAUERSTOFFZUFUHR WURDE ENTWICKELT,
UM DIE AUSBREITUNG VON FEUER ZUR
SICHERHEIT DES PATIENTEN ZU VERRINGERN.
DIE VERWENDUNG NICHT ZUGELASSENER
ZUBEHÖRTEILE KANN DAS VERLETZUNGSODER SCHADENSRISIKO IM ZUSAMMENHANG
MIT BRÄNDEN ERHÖHEN.
WARNUNG: DER SAUERSTOFFKONZENTRATORMUSS AUSGESCHALTET UND VON DER
EXTERNEN STROMVERSORGUNG GETRENNT
SEIN, BEVOR SIE DEN AKKU ENTNEHMEN.
WARNUNG: KLEINE TEILEKÖNNEN EINE POTENZIELLEERSTICKUNGSGEFAHR DARSTELLEN.
WARNUNG: VERMEIDEN SIE DEN KONTAKTMIT DER ABLUFTÖFFNUNG FÜR LÄNGER ALS
10 SEKUNDEN, DA DIESE WARM SEIN KANN.
WARNUNG: WIND ODER STARKE ZUGLUFTKÖNNEN DIE PRÄZISE DURCHFÜHRUNG DERSAUERSTOFFTHERAPIE BEEINTRÄCHTIGEN.
WARNUNG: DURCH DIESES PRODUKT KÖNNEN
SIE MIT CHEMIKALIEN EINSCHLIESSLICH NICKELIN KONTAKT KOMMEN, WELCHES IM STAATKALIFORNIEN ALS KREBSVERURSACHEND GILT.
WEITERE INFORMATIONEN ERHALTEN SIE AUF
WWW.P65WARNINGS.CA.GOV.
78 - DE PN MN232-C4 L | Bedienungsanleitung

VORSICHT: Ziehen Sie immer erst
den Netzstecker aus der Steckdose,
bevor Sie das Netzteil vom
Sauerstoffkonzentrator abziehen.
VORSICHT: Stellen Sie den Sauerstoffkonzentrator
nicht so auf, dass es schwierig ist, an das
Netzkabel zu gelangen.
VORSICHT: Die Sauerstoffversorgungsleitung unddas Stromkabel immer so platzieren, dass niemand
darüber stolpern oder sich damit strangulieren kann.
VORSICHT: Wenn Sie den Sauerstoffkonzentrator
in einem Auto, Boot oder an anderen Gleichstromquellen mit Gleichstromversorgung verwenden,stellen Sie sicher, dass das Fahrzeug gestartet undin Betrieb ist, bevor Sie den Sauerstoffkonzentrator
anschließen. Die Nichtbeachtung dieser Anweisungen kann dazu führen, dass das Netzteil den
Sauerstoffkonzentrator nicht mit Strom versorgt.
VORSICHT: Wenn das Fahrzeug, in dem Sie den
Sauerstoffkonzentrator verwenden, ausgeschaltet
ist, trennen Sie das Gerät vom Fahrzeug und nehmen
Sie es heraus. Lagern Sie den Sauerstoffkonzentrator
nicht in einem sehr heißen oder kalten Auto oderin anderen ähnlichen Umgebungen mit hohen oderniedrigen Temperaturen.
Hinweis: Tragbare und mobile HF-Kommunikationsausrüstung kann medizinische elektrische Geräte
beeinträchtigen.
VORSICHT: Lassen Sie den Sauerstoffkonzentrator oder das Netzteil NICHT im
Fahrzeug eingesteckt, wenn die Zündung
auf AUS gestellt ist. Es kann dabei zu einer
Entladung der Fahrzeugbatterie kommen.
VORSICHT: Der Betrieb oder die Lagerung desSauerstoffkonzentrators außerhalb des normalen
Betriebstemperatur- und Umgebungsdruckbereichs(Höhe) kann die Leistung beeinträchtigen und die
Akkulaufzeit verkürzen und/oder die Akkuladezeit
verlängern. Lagerungs- und Temperaturgrenzwerte
finden Sie im Abschnitt „Technische Daten“ indieser Anleitung.
VORSICHT: Für mit Akkus ausgestattete
Sauerstoffkonzentratoren: Lagern Sie die Akkus kühl
und trocken. Auf diese Weise kann die Lebensdauer
der Akkus verlängert werden. Wenn Sie Ihren
Sauerstoffkonzentrator über einen längerenZeitraum bei hohen Temperaturen oder mit einem
voll aufgeladenen oder vollständig entladenen Akku
aufbewahren, kann sich die Gesamtnutzungsdauer
des Akkus verringern. Versuchen Sie nicht, den
Akku zu öffnen. Er enthält keine wartbaren Teile.
Halten Sie Kinder vom Akku fern.
VORSICHT: VERWENDEN SIE NUR die vom
Hersteller bereitgestellten Akkus. Wenden Sie
sich für Anweisungen zur ordnungsgemäßen
Entsorgung des Akkus an den Bereitsteller Ihres
Gerätes oder eine örtliche zuständige Stelle.

FreeStyle Comfort
Hinweis: Um die durch Konzentrator- oder
Stromausfall verursachten Schäden zu mindern,sollte der Bereitsteller des Geräts während derInstallation und in regelmäßigen Abständen, wennsich diese Eigenschaften ändern, den Bedarf an
Reservesauerstoff beurteilen:

Hinweis: Wenn der Sauerstoffkonzentrator übereinen längeren Zeitraum außerhalb der in der Tabelle
mit den technischen Daten angegebenen normalen
Umgebungsbedingungen gelagert wurde, müssen
Sie ihn vor dem Einschalten auf seine normale
Betriebstemperatur zurückkehren lassen.

- Der Zustand des Patienten

Hinweis: Der Anbieter der Ausrüstung sollte den
Bedarf an Reservesauerstoff für den Fall eines
Konzentrator- oder Stromausfalls bei der Installation
und in regelmäßigen Abständen, wenn sich dieseEigenschaften ändern, prüfen:- Die Umgebung, in der der Patient lebt
- Die Fähigkeit, den Patienten mit Reservelieferungenvon zusätzlichem Sauerstoff zu versorgen.
Hinweis: Sicherstellen, dass die Kanüle vollständig
eingeführt ist und fest sitzt. Damit wird sichergestellt,
dass der Sauerstoffkonzentrator ein Einatmen korrekt
erkennen und Sauerstoff liefern kann. Während desEinatmens sollten Sie den Sauerstofffluss zu den
Stutzen der Nasenkanüle hören oder fühlen können.

- Der Zustand des Patienten- Die Umgebung, in der der Patient lebt
- Die Fähigkeit, den Patienten mit Reservelieferungenvon zusätzlichem Sauerstoff wieder zu versorgen.

Hinweis: Lithium-Ionen-Akkus können dauerhaft an
Kapazität verlieren, wenn sie im vollen oder völlig
entleertem Ladezustand extrem hohen Temperaturen
ausgesetzt sind. Bei längerer Lagerung wird
empfohlen, dass die Akkus zu 25 bis 50 % geladenund bei Temperaturen von -20°C – 20°C(-4°F – 68°F) gelagert werden.
Hinweis: Betreiben Sie den Sauerstoffkonzentrator
nicht ohne eingesetzten Lufteinlassfilter. Sollte einzweiter Filter mitgeliefert worden sein, setzen Sie den
Ersatzfilter ein, bevor Sie den verschmutzten Filter
reinigen. Reinigen Sie den verschmutzten Filter inwarmem Seifenwasser und trocknen Sie ihn vor der
Verwendung gründlich ab.
Hinweis: Der Hersteller empfiehlt, dieses Gerät nicht
zu sterilisieren.
Hinweis: Der Sauerstoff innerhalb eines Raumes
kann nicht aufgebraucht werden, wenn Sie den
Sauerstoffkonzentrator verwenden.
WARNUNG: DER SAUERSTOFFKONZENTRATOR SOLLTE NICHTBETRIEBEN ODER AN EINE EXTERNESTROMQUELLE (WECHSELSTROM/GLEICHSTROM) ANGESCHLOSSEN
WERDEN, WENN DER AKKU NICHT EINGELEGT IST. VERSUCHEN SIE NICHT,DAS GERÄT EINZUSCHALTEN ODER AN
EINE EXTERNE STROMQUELLE ANZUSCHLIESSEN, WENN DER AKKUPACKENTFERNT WURDE.
ENTFERNEN SIE DEN AKKUPACK NUR, WENN DASGERÄT AUSGESCHALTET UND VON DER STROMVERSORGUNG GETRENNT WURDE. VERSUCHENSIE NICHT, DEN AKKUPACK ZU ENTFERNEN, WENNDAS GERÄT NOCH EINGESCHALTET ODER AN DIE
STROMVERSORGUNG ANGESCHLOSSEN IST.
PN MN232-C4 L | Bedienungsanleitung DE- 79

FreeStyle Comfort
Technische Daten
FreeStyle Comfort
Abmessungen (mit Einzelakku)

25,4 x 18,5 x 7,9 cm (10,0 x 7,3 x 3,1 in)

Abmessungen (mit Doppelakku)

27,9 x 18,5 x 7,9 cm (11,0 x 7,3 x 3,1 in)Gewicht (mit Einzelakku)

2,3 kg (5 lbs)

Gewicht (mit Doppelakku)

2,7 kg (6 lbs)

Nenngeräuschpegel***

39,93 dB(A) ± 0,42 dB(A) bei 2

Durchfluss-Einstellungen

Impulseinstellung 1, 2, 3, 4, 5

O2 Ausgabe-Einstellung

Einstellung 1 - 210 ml/min
Einstellung 2 - 420 ml/min
Einstellung 3 - 630 ml/min
Einstellung 4 - 840 ml/min
Einstellung 5 - 1050 ml/min([@20bpm STPD] Alle Volumen +/- 15 %)Sauerstoffkonzentration*

90 % (+5,5 % / -3 %)Sauerstoffmonitor Ja
Maximaler Ausgangsdruck
Dosierempfindlichkeit
Wechselstrom
Gleichstrom< 2 bar (30 psi)
> -0,1 cm H2O (einstellbar)
100–240 VAC, 50–60 Hz, 120 W max.
11–18 V (max. 10 Ampere)

Akkutyp Lithium-Ionen
Akkukapazität
Aufladezeit der Akkus
Laufzeiten der Akkupacks

Einzelzelle: 6700 mAh; Doppelzelle: 13400 mAh
Einzelakku: 3,5 Stunden
Doppelakku: 6 Stunden
Laufzeit Einzelakku: Einstellung 2 - bis zu 4 Stunden
Laufzeit Doppelakku: Einstellung 2 - bis zu 8 Stunden

Betriebstemperatur** 5 °C bis 40 °C (41 °F bis 104 °F)
Transport- und Lagertemperatur. (Einheit)-25 °C bis +70 °C (-13 °F bis +158 °F) bis zu 3048 m (10000 ft)

Transport- und Lagertemperatur. (Batterie)-25 °C bis +70 °C (-13°F to +158°F) bis zu 3048 m (10000 ft)

Betriebshöhe**Relative Luftfeuchtigkeit
Lagerfeuchtigkeit
Beschränkte Garantie-381 bis 3.048 m (-1.250 bis 10.000 ft)(getestet bei 700–1.060 hPa)
15–95 % relative Luftfeuchtigkeit (nicht kondensierend)
0–90 %, nicht kondensierend
Siehe Garantieerklärung

* Werte bei 21 °C (70 °F) und einem Luftdruck von 101 kPa (14,7 psi)** Das Gerät funktioniert in allen Umgebungen, solange die Umgebung den angegebenen Temperatur- undFeuchtigkeitsbereich einhält. Ein Betrieb außerhalb dieser Spezifikationen kann die Fähigkeit des Konzentrators
einschränken, die Spezifikation der Sauerstoffkonzentration bei höheren Literdurchflussmengen zu erfüllen.
Der Aufwärmbildschirm kann für bis zu 15 Minuten angezeigt werden oder bis der Konzentrator die Zielkonzentration
erreicht hat. Das Gerät kann während dieser Zeit weiterhin verwendet werden.
*** Schallpegel gemessen nach Prüfmethode Nr. 14-1 10/2018 MDS-Hi.
Die voraussichtliche Nutzungsdauer beträgt mindestens fünf Jahre.

80 - DE PN MN232-C4 L | Bedienungsanleitung

FreeStyle Comfort
Gebrauchsanweisungen
1. Platzieren und positionieren Sie den FreeStyle
Comfort an einem für die Nutzung geeigneten
Standort.
a. Der Konzentrator sollte gut belüftet sein,
damit die Lufteinlässe und -auslässe nicht
blockiert werden.
b. Der Konzentrator sollte so positioniert
werden, dass alle akustischen und optischen
Anzeigen und Alarme leicht gehört undgesehen werden können.
c. Vergewissern Sie sich, dass die Lufteinlassfilter
angebracht sind, bevor Sie das Gerät in Betrieb
nehmen. Wenn für das Gerät ein Filter fehlt,
wenden Sie sich bitte an Ihren Geräteanbieter.

3. Stecken Sie das Wechselstrom- oder
Gleichstrom-Netzteil in den Eingang für den
Stromanschluss unten rechts an der Vorderseite
des Geräts.

2. Legen Sie einen Akkupack in den Konzentrator
ein.
a. R
 ichten Sie den Akku mittig auf die Unterseite
des Gehäuses Ihres FreeStyle Comfort aus.

4. Schließen Sie eine Nasenkanüle an die Spitze
des Sauerstoffauslasses an. Vergewissern
Sie sich vor der Inbetriebnahme, dass die
Verbindung stabil ist.

b. Schieben Sie den Akku nach innen, bis Sie
ein Klicken hören, das bestätigt, dass er
eingerastet ist.

HINWEIS: Das Verwenden von anderen als
den empfohlenen Kanülen kann die Leistung
beeinträchtigen.

Hinweis: An den Geräteanbieter: Das folgende
Zubehör für die Sauerstoffverabreichung wird für die
Verwendung mit dem FreeStyle Comfort empfohlen:• Nasenkanüle mit 2,1 m (7 ft) langem Schlauch(Max. l/min: 6 l/min): CAIRE Teilenummer 5408-SEQ• Weitere Informationen zu empfohlenem Zubehör
sind online unter www.caireinc.com verfügbar.
5. Schalten Sie den FreeStyle Comfort ein, indem
Sie den Ein-/Aus-Schalter drücken. Wählen
Sie die Durchflusseinstellung, die Ihnen
verschrieben wurde, indem Sie auf Erhöhen (+)
oder Reduzieren (-) drücken.

PN MN232-C4 L | Bedienungsanleitung DE- 81

FreeStyle Comfort

Ein kurzer akustischer Alarm ertönt, um
anzuzeigen, dass das Gerät eingeschaltet ist.
Hinweis: Wenn der FreeStyle Comfort zum ersten
Mal eingeschaltet wird, ist standardmäßig die
Durchfluss-Einstellung 2 eingestellt. Wenn die
Einstellung geändert wird, schaltet sich das Gerät auf
die vorherige Durchflusseinstellung ein.
6. Positionieren Sie die Nasenkanüle entsprechend
der Anleitung des Herstellers, die der Kanüle
beiliegt, in Ihrem Gesicht und atmen Sie normal.
Bei jedem Einatmen müssten Sie spüren, wie ein
Sauerstoffstrom durch die Nasenkanüle fließt.
7. Um den Durchfluss auf die gewünschte
Einstellung zu bringen, drücken Sie auf
Erhöhen (+) oder Reduzieren (-). Es ist normal,
dass sich das Geräusch bei Änderung derDurchflusseinstellungen ändert.

Atemfrequenz
15
20
25
30
35
40

autoSAT Impulsvolumen (ml)*Einstellung
1
2
3
4
5
10,5
10,5
10,5
10,5
10,5
10,5

21,0
21,0
21,0
21,0
21,0
21,0

31,5
31,5
31,5
31,5
30,0
26,3

42,0
42,0
42,0
35,0
30,0
26,3

*Toleranz ist +/- 15 %82 - DE PN MN232-C4 L | Bedienungsanleitung

52,5
52,5
42,0
35,0
30,0
26,3

nicht-autoSAT Impulsvolumen (ml)*Einstellung
Atemfre- 1
2
3
4
5
quenz
15
14,0 28,0
42,0
56,0
70,0
20
10,5 21,0
31,5
42,0
52,5
25
8,4
16,8
25,2
33,6
42,0
30
7,0
14,0
21,0
28,0
35,0
35
6,0
12,0
18,0
24,0
30,0
40
5,3
10,5
15,8
21,0
26,3
8. Schalten Sie den FreeStyle Comfort aus, indem
Sie den Netzschalter drücken.
Hinweis: Der Bildschirm wechselt in den StandbyModus und wird nach 20 Sekunden abgedunkelt.
Drücken Sie eine beliebige Taste, um den Bildschirm
aus dem Standby-Modus zu wecken.
9. Für erweiterte Funktionen siehe Anweisungenzur Menünavigation.

SPEZIELLE FUNKTIONENDES PRODUKTS
autoSAT®
CAIREs proprietäre AutoSAT-Funktion ermöglicht
es dem FreeStyle Comfort, ein konstantes
Sauerstoffvolumen (Bolus) pro Atemzug bis
zu einer maximalen Atemfrequenz zu liefern.
Wenn sich die Atemfrequenz des Benutzers bei
Aktivitäten wie Gehen erhöht, passt die autoSATFunktion den Gerätebetrieb an, um ein eingestelltes
Bolusvolumen beizubehalten.
UltraSense®
Dieser empfindliche Atemimpuls-Trigger erkennt
Druckveränderungen während des Atmens undführt Sauerstoff beim Einatmen zu.
autoDOSE®
Falls ein Apnoe-Alarm (Atempausen-Alarm)
auftritt, führt diese Funktion bei 20 Atemzügen
pro Minute Sauerstoff in der am Gerät eingestellten
Durchflussrate zu, bis die Alarmbedingung
behoben ist.

FreeStyle Comfort
Stromversorgung
Der FreeStyle Comfort kann auf 3 Arten betrieben
werden:• Wiederaufladbarer Akkupack• Wechselstrom• Gleichstrom
Informationen zum Betrieb dieser Komponenten
finden Sie in den folgenden Abschnitten.
WARNUNG: VERWENDEN SIE NUR
DIE IN DER BEDIENUNGSANLEITUNG
AUFGEFÜHRTEN STROMQUELLEN. BEI
DER VERWENDUNG VON NETZTEILEN
UND AKKUPACKS, DIE NICHT VONCAIRE HERGESTELLT WURDEN, KANNDAS GERÄT BESCHÄDIGT ODER DER
BETRIEB BEEINTRÄCHTIGT WERDEN.

ENTFERNEN DES AKKUS

Drücken Sie den Akku-Entriegelungsknopf nach
unten und weg vom FreeStyle Comfort. HALTEN
Sie diesen Knopf gedrückt, während Sie den
Akku durch Wegdrücken vom Konzentrator
herausnehmen.
WARNUNG: DER SAUERSTOFFKONZENTRATOR MUSS VOR DEM ENTFERNEN
DES AKKUPACKS AUSGESCHALTETUND VON DER EXTERNEN STROMVERSORGUNG GETRENNT WERDEN.

AKKUPACK

Der FreeStyle Comfort kann mit dem mitgelieferten
wiederaufladbaren Akkupack betrieben werden.
Wird der Konzentrator mit dem Akkupack
verwendet, gibt es keine Einschränkungender Funktion. Bei Akkubetrieb sind alle
Durchflussraten verfügbar.

ÜBERPRÜFEN DES AKKUPACKLADEZUSTANDSEINSETZEN DES AKKUS

Richten Sie den Akku mittig auf die Unterseite desGehäuses Ihres FreeStyle Comfort aus. Drücken
Sie den Akku hinein, bis er mit einem Klicken
einrastet.

Wenn Sie den Ladestand des Akkus überprüfen
möchten, während dieser im FreeStyle Comfort
eingelegt ist, sehen Sie sich das Batteriesymbol unddie geschätzte verbleibende Zeit auf dem Display an.
80–100 %60–80 %

40–60 %

20–40 %20 %PN MN232-C4 L | Bedienungsanleitung DE- 83

FreeStyle Comfort
Hinweis: Wenn das Netzteil angeschlossen ist, wird
anstelle der verbleibenden Ladung ein Steckersymbol
eingeblendet.
Wenn Sie den Akku-Ladezustand überprüfen
möchten, während sich der Akku nicht im FreeStyle
Comfort befindet, legen Sie den Akku flach hin
und drücken Sie die TEST-Taste. Die AkkuLadeleuchten (25–100 %) leuchten rechts neben
der TEST-Taste auf und geben den Ladezustand des
Akkupacks an.

AKKU-LAUFZEIT

Wird die externe Wechsel- oder
Gleichstromversorgung unterbrochen, schaltet derKonzentrator automatisch auf Akkubetrieb, wenn
der Akkupack eingelegt und aufgeladen ist. Die
folgende Tabelle zeigt die typischen Laufzeiten
eines neuen Akkupacks. Die Akkulaufzeit bleibt
unabhängig von der Atemfrequenz konstant.
Impulseinstellung

Laufzeit desEinzelakkus(8-Zellen)

Laufzeit desDoppelakkus(16-Zellen)

1

bis zu 8 Stunden

bis zu 16 Stunden2

bis zu 4 Stunden

bis zu 8 Stunden3

bis zu 3 Stunden

bis zu 6 Stunden4

bis zu 2,25 Stunden

bis zu 4,5 Stunden

5

bis zu 2 Stunden

bis zu 4 Stunden

TYPISCHE AKKU-LADEZEITANFÄNGLICHE LADUNG

Der neue Akku Ihres FreeStyle Comfort ist bei
Auslieferung nur teilweise geladen. Vor derersten Verwendung sollten Sie den Akkupack mit
Wechsel- oder Gleichstrom vollständig aufladen.

ALLGEMEINE INFORMATIONEN ZUM
AUFLADEN

Der FreeStyle Comfort lädt den Akkupack immer
dann auf, wenn er sich im Konzentrator befindet
und der Konzentrator an eine externe Wechsel- oder
Gleichstromquelle angeschlossen ist (außer im
Flugzeug).
Der Akku ist vollständig geladen, wenn er 100 %anzeigt.
Alternativ kann der Akkupack mit dem optionalen
Tischladegerät geladen werden.

84 - DE PN MN232-C4 L | Bedienungsanleitung

Die benötigte Zeit zum vollständigen Aufladeneines vollständig entladenen Akkus beträgt
3,5 Stunden für einen Einzelakku und 6 Stundenfür einen Doppelakku.
Hinweis: Wenn der Akku vollständig entladen ist,
beginnt der Ladevorgang möglicherweise nicht sofort.

AKKU-LEBENSDAUER

Der Akku des FreeStyle Comfort ist für
300 Lade- und Entladezyklen ausgelegt.

FreeStyle Comfort
Wechselstrom- undGleichstromversorgung
Der FreeStyle Comfort wird standardmäßig mit
einem Wechsel- und einem Gleichstromnetzteil
geliefert. Bestandteile der Wechselstromversorgung:	a. Wechselstromnetzteil für den Betrieb mit
100–240 V, 50–60 Hz.
	b. Wechselstromkabel zum Anschluss desNetzteils an die entsprechende Steckdose.

Beim Gleichstromnetzteil handelt es sich um
ein Kabel, das direkt an das Gerät und an die
Gleichstromsteckdose angeschlossen wird.

2. Verbinden Sie das Wechselstromkabel mit dem
Wechselstromnetzteil.

WECHSELSTROMVERSORGUNG

Das Wechselstromnetzteil ist für den Hausgebrauchoder ähnliche Umgebungen mit standardmäßigen
Wechselstromsteckdosen vorgesehen.
Der FreeStyle Comfort ist mit Wechselstromvoll funktionsfähig. Der FreeStyle Comfort lädt
den Akkupack zudem immer dann auf, wenn
Wechselstrom zur Verfügung steht(außer im Flugzeug). Die Aufladung des Akkupacks
erfolgt unabhängig davon, ob der FreeStyle
Comfort ein- oder ausgeschaltet ist.
WARNUNG: SCHLIESSEN SIE KEINE
EXTERNE STROMVERSORGUNG AN,
WENN SICH DER AKKUPACK NICHTIM SAUERSTOFFKONZENTRATOR
BEFINDET.

3. Schließen Sie das Steckerende desWechselstromkabels an eine Wandsteckdoseoder eine andere geeignete Steckdose an.

Um den FreeStyle Comfort an die
Wechselstromversorgung anzuschließen:
1. Verbinden Sie das Wechselstromnetzteil mit
der Netzanschlussbuchse unten rechts an der
Vorderseite des FreeStyle Comfort. Führen Sie
den Stecker bis zum Anschlag ein, um einesichere Stromverbindung zu gewährleisten.

PN MN232-C4 L | Bedienungsanleitung DE- 85

FreeStyle Comfort
4. Schalten Sie den FreeStyle Comfort ein, indem
Sie den Ein-/Aus-Schalter drücken. Wählen
Sie die Durchfluss-Einstellung, die Ihnen
verschrieben wurde, indem Sie auf Erhöhen (+)
oder Reduzieren (-) drücken.

Hinweis: Wenn der Akkupack wieder aufgeladen
werden muss, wird das Batteriesymbol auf dem
Display mit einem orangefarbenen Balken angezeigt.

GLEICHSTROM

Für den Betrieb über Steckdosen in Autos, Booten
und anderen Kraftfahrzeugen benötigen Sie das
Gleichstromnetzteil. Der FreeStyle Comfort ist mit
Gleichstrom voll funktionsfähig. Der FreeStyle
Comfort lädt den Akkupack bei Bedarf wieder
auf, wenn Gleichstrom zur Verfügung steht. Der
Akkupack wird sowohl bei ausgeschaltetem
Gerät als auch bei eingeschaltetem und in Betrieb
befindlichem Gerät aufgeladen.
WARNUNG: SCHLIESSEN SIE KEINE
EXTERNE STROMVERSORGUNG
AN, WENN KEIN AKKUPACK IMSAUERSTOFFKONZENTRATORINSTALLIERT IST.
1. Starten Sie Ihr Kraftfahrzeug.
2. Verbinden Sie das Gleichstromnetzteil mit dem
Eingang für den Stromanschluss unten rechts an
der Vorderseite des FreeStyle Comfort. Führen
Sie den Stecker bis zum Anschlag ein, um einesichere Stromverbindung zu gewährleisten.

86 - DE PN MN232-C4 L | Bedienungsanleitung

3. Schließen Sie das Steckerende des GleichstromNetzkabels an die Autosteckdose oder an eine
andere geeignete Gleichstromquelle an.

4. Schalten Sie den FreeStyle Comfort ein, indem
Sie den Ein-/Aus-Schalter drücken. Wählen
Sie die Durchfluss-Einstellung, die Ihnen
verschrieben wurde, indem Sie auf Erhöhen (+)
oder Reduzieren (-) drücken.
VORSICHT: Um ein mögliches Feuer odereine Überhitzung zu vermeiden, stellen
Sie sicher, dass die 12-V-DC-Steckdose
frei von Fremdkörpern ist und derAdapterstecker des DC-Netzteils richtig
sitzt und sicher an Ort und Stelle bleibt.
VORSICHT: Metalloberflächen vonGleichstromversorgungen können durch den
hohen Strom SEHR HEISS werden, wenn sie in
Betrieb sind. Vor dem Berühren ausreichend Zeit
zum Abkühlen lassen.
VORSICHT: Verwenden Sie das Gleichstromnetzteil
nicht mit Verlängerungen, Splittern oder Adaptern.
Die Verwendung nicht zugelassener Zubehörteile
kann zu einer Überhitzung oder Beschädigung
führen.
VORSICHT: Das Anschließen des Geräts an eine
12-V-DC-Steckdose kann die Fahrzeugbatterieentladen, wenn der Motor nicht läuft, auch wenndas Gerät ausgeschaltet ist.
VORSICHT: Führen Sie keine Starthilfe für ein
Fahrzeug durch, während das Gleichstromnetzteil
angeschlossen ist.

FreeStyle Comfort
Reinigung, Pflege und regelmäßige
Wartung
REGELMÄSSIGE WARTUNG

Vorbeugende Wartung
Das Gerät ist wartungsfrei. Filter im Inneren desGeräts sind für die Nutzungsdauer des Geräts
ausgelegt. Wenn das Gerät nicht ordnungsgemäß
funktioniert, sollte es zur Reparatur an den Anbieter
zurückgegeben werden. Wartungsarbeiten an deninternen Komponenten des FreeStyle Comfort müssen
vom Sauerstoff-Versorger durchgeführt werden.
Der Betreiber des Geräts ist für das folgende
Wartungsverfahren verantwortlich.
• Wöchentlich – Reinigung der Luftansaugfilter
WARNUNG: VERSUCHEN SIE NICHT,DIE KUNSTSTOFFABDECKUNGENDES SAUERSTOFFKONZENTRATORSZU ÖFFNEN ODER ZU ENTFERNEN.
DAS GERÄT SOLLTE NUR VON EINEMQUALIFIZIERTEN SERVICETECHNIKER
GEÖFFNET WERDEN.

LUFTANSAUGFILTER

Der FreeStyle Comfort nimmt Luft über die
Luftansaugfilter auf, die sich unten auf der
Vorderseite des Sauerstoffkonzentrators befinden.
Diese Filter verhindern das Eindringen von Staub
und anderen größeren Partikeln in das Gerät.
Die Luftansaugfilter des Geräts sollten mindestenseinmal pro Woche gereinigt werden. Ihr SauerstoffVersorger kann Ihnen, je nach Einsatzbedingungendes Geräts, eine häufigere Filterreinigung
empfehlen. Führen Sie die folgenden Schritte aus,
um die Luftansaugfilter sachgemäß zu reinigen.
1. Entfernen Sie die beiden Filter von der
Vorderseite des FreeStyle Comfort, indem Sie
die entsprechende Lasche nach oben drücken
und nach außen ziehen.

2. Reinigen Sie beide Filter in einer Lösung aus
warmem Wasser und Seife.
3. Spülen Sie die Filter gründlich aus und entfernen
Sie überschüssiges Wasser mit einem weichen,fusselfreien Tuch. Stellen Sie vor Wiedereinbau
sicher, dass die Filter vollständig trocken sind.

4. Setzen Sie die sauberen und trockenen Filter
wieder ein.
WARNUNG: BETREIBEN SIE DEN SAUERSTOFFKONZENTRATOR NICHT OHNE
EINGESETZTEN LUFTEINLASSFILTER.

RICHTIGER UMGANG MIT DEM AKKUPACK
Vermeiden Sie, dass der Akkupack mit
Flüssigkeiten in Berührung kommt.

Versuchen Sie nicht, den Akkupack zu öffnen oderzu zerlegen.
Setzen Sie den Akkupack keinen offenen
Flammen aus.
Lassen Sie den Akkupack nicht fallen.
Bewahren Sie den Akkupack außerhalb
der Reichweite von Kindern auf.
Wenn Ihr Akkupack beschädigt erscheint,
kontaktieren Sie Ihren Sauerstoff-Versorger
vor einem erneuten Gebrauch.
Halten Sie den Akkupack von Mikrowellen undanderen Kochgeräten fern.

LAGERUNG DES AKKUPACKS

Der Akkupack des Sauerstoffkonzentrators muss
vor dem Aufladen nicht vollständig entladen
werden. Es wird empfohlen, den Akkupack desSauerstoffkonzentrators nach jedem Gebrauch
aufzuladen.
Lagern Sie den Akkupack nicht an Orten, an denen
er starker Hitze ausgesetzt ist, wie z. B. in direktem
Sonnenlicht oder im Auto. Andernfalls können
Leistung und Lebensdauer des Akkupacks starkbeeinträchtigt werden. Im schlimmsten Fall kannder Akkupack Hitze und Rauch entwickeln und
ggf. Feuer fangen oder explodieren. Der Akkupack
sollte in einer kontrollierten Umgebung mit einer
relativen Luftfeuchtigkeit unter 70 % und frei
von korrosiven Gasen gelagert werden. Um Rostvorzubeugen, vermeiden Sie Bedingungen, die
Kondensation verursachen können, wie zum
Beispiel große Temperaturschwankungen.
Für eine Lagerungszeit von einem Monatoder mehr wird empfohlen, den Akkupack im
vollständig entladenen oder teilweise geladenen(50 % bis 75 % – zwei bis drei LEDs) Zustandzu lagern. Entfernen Sie den Akkupack aus dem
Sauerstoffkonzentrator, sobald er vollständig
geladen ist.
PN MN232-C4 L | Bedienungsanleitung DE- 87

FreeStyle Comfort
REINIGUNG

Schalten Sie das Gerät vor der Reinigung aus
und trennen Sie es von der Wechsel- oder
Gleichstromversorgung. Sprühen Sie das Gehäuse
NICHT direkt ein. Verwenden Sie ein feuchtes(nicht nasses) Tuch oder einen Schwamm.
Besprühen Sie das Tuch oder den Schwamm mit
einer milden Haushaltsreinigerlösung, um das
Gehäuse und die Netzteile zu reinigen. Gehen Sie
gemäß den Herstelleranweisungen vor.
WARNUNG: ZIEHEN SIE DASNETZKABEL AUS DER STECKDOSE,
BEVOR DAS GERÄT GEREINIGTWIRD, UM DIE GEFAHR EINESSTROMSCHLAGS ODER EINERVERBRENNUNG AUSZUSCHLIESSEN.
NUR DER GERÄTEANBIETER ODER EINQUALIFIZIERTER SERVICETECHNIKER
DÜRFEN DIE ABDECKUNG ENTFERNENODER DAS GERÄT WARTEN.
WARNUNG: ES MUSS DARAUF GEACHTET WERDEN, DASS DER SAUERSTOFFKONZENTRATORSTETS TROCKEN BLEIBT UND DASS KEINE FLÜSSIGKEIT IN DAS GERÄT GELANGT. ANDERNFALLSKANN EINE FEHLFUNKTION AUFTRETEN ODEREINE ABSCHALTUNG ERFOLGEN UND ES BESTEHT
EIN ERHÖHTES RISIKO FÜR EINEN ELEKTRISCHENSCHLAG UND FÜR VERBRENNUNGEN.
WARNUNG: SETZEN SIE DIREKT AUF DEM GERÄTKEINE FLÜSSIGKEITEN EIN. DIE LISTE DER
UNERWÜNSCHTEN CHEMISCHEN WIRKSTOFFE
BEINHALTET FOLGENDE FLÜSSIGKEITEN,IST JEDOCH NICHT AUF DIESE BESCHRÄNKT:
ALKOHOL UND ALKOHOLISCHE PRODUKTE,KONZENTRIERTE CHLORHALTIGE PRODUKTE(ETHYLENCHLORID) UND PRODUKTE AUF
ÖLBASIS. EINE REINIGUNG MIT DIESENPRODUKTEN KANN DAS PLASTIKGEHÄUSE
NACHHALTIG BESCHÄDIGEN.
WARNUNG: DAS GEHÄUSE, BEDIENFELD UND
NETZKABEL MIT EINEM SCHONENDEN HAUSHALTSREINIGER REINIGEN. DAZU EIN FEUCHTES(ABER KEIN NASSES) TUCH ODER EINEN FEUCHTEN SCHWAMM VERWENDEN. ALLE BESTANDTEILE NACH DER REINIGUNG ABTROCKNEN. KEINEFLÜSSIGKEIT IN DAS GERÄT EINDRINGEN LASSEN. BESONDERS DARAUF ACHTEN, DASS DER
SAUERSTOFFAUSLASS FÜR DEN ANSCHLUSSDER KANÜLE FREI VON STAUB, WASSER UND
ANDEREN PARTIKELN BLEIBT.

88 - DE PN MN232-C4 L | Bedienungsanleitung

WARNUNG: VERWENDEN SIE EIN FEUCHTES(ABER KEIN NASSES) TUCH ODER EINENFEUCHTEN SCHWAMM, UM DEN AKKUPACK ZUREINIGEN. SPRÜHEN SIE DAS TUCH BZW. DEN
SCHWAMM ERST MIT EINEM SCHONENDEN
HAUSHALTSREINIGER EIN UND REINIGEN SIE
ANSCHLIESSEND DEN AKKUPACK.
WARNUNG: REINIGEN SIE DAS GERÄT NICHT MIT
EINEM AUTOMATISCHEN REINIGUNGSSYSTEM.

ERSETZEN DER KANÜLE

Für einen ordnungsgemäßen Gebrauch stets den
Anleitungen des Kanülenherstellers folgen. Ersetzen
Sie die Nasenkanüle oder den Sauerstoffschlauchalle 3 bis 6 Monate (bei täglicher Anwendung alle
3 Monate) entsprechend den Empfehlungen desKanülenherstellers oder Ihres Sauerstoff-Versorgers.
Informationen zur Reinigung und zum Austauscherhalten Sie von Ihrem Arzt oder Pflegedienst.
Weitere Ersatzteile erhalten Sie von Ihrem
Sauerstoff-Versorger.

FreeStyle Comfort
Warnzustände
Prioritätsnummer
1

2

Warnbeschreibung
Systemstörung Hohe
Akku-Temperatur

Systemstörung hohe
Temperatur

Anzeige

Akustisches
Signal1 Signaltonalle
20 SekundenWeitere Informationen / Empfohlene
Maßnahme
Das Gerät warnt, wenn sich die Temperatur
des Akkus dem Grenzwert für die Sicherheitsabschaltung nähert.
Wenn die Abschaltschwelle erreicht ist,schaltet das Gerät ab. Wenn eine externe
Stromversorgung angeschlossen ist, wird
der Warnbildschirm angezeigt und das
akustische Warnsignal wird für 2 Minuten
ausgegeben.

1 Signaltonalle
20 Sekunden

Testen Sie mit dem Akku bei Zimmertemperatur. Wenn das Problem weiterbesteht,
wenden Sie sich an den Geräteanbieter.
Das Gerät warnt, wenn sich die interne
Temperatur dem Grenzwert für die Sicherheitsabschaltung nähert.
Wenn die Abschaltschwelle erreicht ist,schaltet sich das Gerät ab, der Warnbildschirm wird angezeigt und das akustische
Warnsignal wird für 2 Minuten ausgegeben.

3

4

5

Systemstörung
hoher Druck(Hoher
Systemdruck- oder
Kompressorfehler)
Systemstörung Durchflussmenge(Ausfall des
OCD-Ventils, Blockierung desGasweges)
Abschaltung
bei schwachem Akku

1 Signaltonalle
20 Sekunden

Dafür sorgen, dass die Abluftöffnungendes Geräts nicht blockiert sind. Wenn das
Problem weiterbesteht, wenden Sie sich an
den Geräteanbieter.
Wenn die Abschaltschwelle erreicht ist,schaltet sich das Gerät ab, der Warnbildschirm wird angezeigt und das akustische
Warnsignal wird für 2 Minuten ausgegeben.
Wenn das Problem weiterbesteht, wenden
Sie sich an den Geräteanbieter.

1 Signaltonalle
30 Sekunden

Wenn das Problem weiterbesteht, wenden
Sie sich an den Geräteanbieter.

1 Signaltonalle
30 Sekunden

Warnmeldungen bis der Akku leer ist unddas Gerät heruntergefahren wird.
Laden Sie den Akku auf.

PN MN232-C4 L | Bedienungsanleitung DE- 89

FreeStyle Comfort
Prioritätsnummer
6

Warnbeschreibung
Keine
Atmungerkannt

Anzeige

Akustisches
Signal1 Signaltonalle
10 Minuten

Weitere Einzelheiten
Es wurde 15 Sekunden lang kein Atemzug
festgestellt.
Wird nach dem ersten Atemzug oder nachder Abschaltung des Aufwärmbildschirms
aktiviert, je nachdem, was früher eintritt.
Überprüfen Sie die Kanüle auf Knickeoder Undichtigkeiten. Erhöhen Sie die
Empfindlichkeitseinstellungen. Wenn das
Problem weiterbesteht, wenden Sie sich an
den Geräteanbieter.

7

Aufwärmen

KeinesDas Gerät beginnt mit der automatischen
Dosierung, bis ein Atemzug erkannt wirdoder das Aufwärmen endet, je nachdem,
was früher eintritt.

8

Niedrige
Konzentration

1 Signaltonalle
10 Minuten,
bis zu30 Minuten
lang

Wenn das Problem weiterbesteht, wenden
Sie sich an den Geräteanbieter.

9

Warnung bei
schwachem
Akku

1 Signalton,keine Wiederholung

Laden Sie den Akku auf.

10

Atemfrequenz überschritten

1 Signalton,keine Wiederholung

Atemfrequenz übersteigt 40 Atemzüge pro
Minute.

11

Systemstörung
Lüfterausfall

Keines12

Systemstörung
niedriger
Druck

KeinesWenn das Problem weiterbesteht, wenden
Sie sich an den Geräteanbieter.

13

Systemstörung
niedrige
Temperatur

KeinesGeben Sie dem Gerät Zeit, sich auf Zimmertemperatur zu erwärmen. Wenn das
Problem weiterbesteht, wenden Sie sich an
den Geräteanbieter.

90 - DE PN MN232-C4 L | Bedienungsanleitung

Wird eine falsche Dosis abgegeben,
reduzieren Sie die Empfindlichkeitseinstellungen. Wenn das Problem weiterbesteht,
wenden Sie sich an den Geräteanbieter.
Wenn das Problem weiterbesteht, wenden
Sie sich an den Geräteanbieter.

FreeStyle Comfort
Prioritätsnummer
Info

Warnbeschreibung
Inbetriebnahme

Anzeige

Akustisches
Signal
1 Info-Signalton

Weitere Einzelheiten
Der Startbildschirm wird beim Einschaltenangezeigt und Sie sollten einen einzelnen
Signalton hören, der die Funktion desSignalgebers bestätigt.
Der Startbildschirm zeigt die aktuelle Firmware-Version in der rechten unteren Ecke an.

InfoNormalanzeige mit
autoSATModus(nicht aneine Gleichstromquelleangeschlossen)
Normalanzeige mit
autoSAT
deaktiviert(nicht aneine Gleichstromquelleangeschlossen)
Einstellung
geändert

KeinesInfoExterne
Stromversorgung imausgeschalteten
Zustandeinstecken

KeinesInfoExterne
Stromversorgung im
Betriebszustandeinstecken

KeinesInfoExterne
Stromversorgung imausgeschalteten
Zustandtrennen1 Info-Signalton

Info

Info

Keines1 Info-SignaltonAnzeige des CAIRE-Bildschirms für 2 Sekunden, gefolgt von Akku- und Steckerstatus für 5 Sekunden, bevor der Bildschirm
ausgeschaltet wird.

Anzeige des Caire-Bildschirms für
2 Sekunden und anschließend Abschaltung
der Anzeige.

PN MN232-C4 L | Bedienungsanleitung DE- 91

FreeStyle Comfort
Prioritätsnummer
Info

Info

Info

Info

Warnbeschreibung
Externe
Stromversorgung im
Betriebszustandtrennen(bei vorhandenem
Akku)
Im Standby-Modus –wenn dieAufwärtsoder
Abwärtstaste gedrückt
wird, zeigt
der Bildschirm den
Akku- undLadezustand an
FlugzeugModus bei
Akkubetrieb

Anzeige

Flugzeug-Modus
bei Netzbetrieb

92 - DE PN MN232-C4 L | Bedienungsanleitung

Akustisches
Signal
1 Info-Signalton

Weitere Einzelheiten

KeinesDer Bildschirm wird 5 Sekunden langangezeigt und dann ausgeschaltet.

1 langer
Signalton

1 langer
Signalton

FreeStyle Comfort
Anweisungen zur Menüführung
Sie können auf das Menü im Standby- oder im
Betriebsmodus zugreifen. Tritt während desBetriebsmodus ein Alarmzustand auf, wird das
Menü automatisch verlassen. Wird innerhalb
20 Sekunden keine Taste gedrückt, wird das Menü
automatisch verlassen. Um das Gerät aus demStandby-Modus aufzuwecken, müssen Sie die
Menü-Taste eventuell zweimal drücken. Dannkönnen Sie auf das Menü zugreifen.
Flugzeug-Modus

Das Menü verfügt über vier Bildschirme:
1. Statusbildschirm: Zeigt die Firmware-Versionsnummer, Laufzeit, Akku-Entladezyklen an
2. Flugzeugmodus (EIN/AUS)
3. Empfindlichkeit: einstellbar (1–5)
4. autoSAT (EIN/AUS)
Empfindlichkeit

Beispiel Status-BildschirmMENÜEINSTELLUNGEN BETRIEB

autoSAT

Drücken Sie im Betriebs- oder Standbymodus
auf die Taste Menü/Auswählen
Menüeinstellungen zuzugreifen.
Drücken Sie, um auf die, um durch die Menüs zu scrollen.

Status-Seite

PN MN232-C4 L | Bedienungsanleitung DE- 93

FreeStyle Comfort
STATUS-SEITEDie Statusseite des Freestyle Comfort zeigt die
Firmware-Versionsnummer, Laufzeit und Anzahl
der Entladezyklen des angeschlossenen Akkus an.

Drücken Sie, während das Symbol LÖSCHEN
angezeigt wird, um das Einschalten des FlugzeugModus abzubrechen.

Flugzeug-Modus Ein Bestätigt
Status-Seite

Laufstunden werden in ganzen Stunden angezeigt.
Akkuentladung ist die Anzahl der Zyklen, wie
oft der angeschlossene Akku entladen und wieder
aufgeladen wurde.

FLUGZEUG-MODUS
Der FreeStyle Comfort wird mit ausgeschaltetem
Flugzeug-Modus geliefert.

Wiederholen Sie diese Vorgänge und wechseln Sie
auf AUS, um den Flugzeug-Modus auszuschalten.
drücken, wenn auf dem Bildschirm
Indem Siedas Symbol „Weiter“ angezeigt wird, wechselt der
Bildschirm zur nächsten Menü-Option.
Der Flugzeug-Modus ist für den Gebrauch im
Betriebsmodus bestimmt. Sobald das Gerät sich imStandby-Modus oder Akkubetrieb befindet, wird
der Flugzeug-Modus automatisch deaktiviert.
Hinweis: Unter Flugzeug-Modus Aktivieren finden
Sie weitere Informationen zum Flugzeug-Modus.

EMPFINDLICHKEITS-MODUS

Drücken Sie, während das Symbol WECHSEL
angezeigt wird, um den Flugzeug-Moduszu aktivieren. Der Wert wechselt auf „EIN“in blauer Farbe.

Der FreeStyle Comfort wird vom Hersteller mit
einer standardmäßigen Empfindlichkeitseinstellung
von 2 geliefert und unter normalen Bedingungensollte keine andere Einstellung erforderlich
sein. Die Empfindlichkeitseinstellung reichtvon den Werten 1–5, wobei 1 für die geringste
Empfindlichkeit steht und 5 für die höchste
Empfindlichkeit.

Flugzeugmodus Ein Ausgewählt

Empfindlichkeitsbildschirm bei Einstellungswert 2

Hauptbildschirm Flugzeug-Modus

Drücken Sie, während das Symbol
SPEICHERN angezeigt wird, um das Einschalten
des Flugzeug-Modus zu bestätigen. Der Wert
wechselt auf „EIN“ in schwarzer Farbe.
94 - DE PN MN232-C4 L | Bedienungsanleitung

FreeStyle Comfort
Drücken Sie, während das Symbol WECHSEL
angezeigt wird, um den Empfindlichkeits-Modus
einzustellen. Der Wert wechselt zur nächsten
blauen Zahl.

Empfindlichkeits-Modus Einstellungswert 3 Ausgewählt

Drücken Sie, während das Symbol
SPEICHERN angezeigt wird, um den
Einstellungswert 3 zu bestätigen. Der Wert
wechselt auf die Zahl „3“ in schwarzer Farbe.

AUTOSAT-MODUS
Der FreeStyle Comfort wird mit eingeschaltetem
autoSAT-Modus geliefert.

Hauptbildschirm autoSAT-Modus

Drücken Sie, während das Symbol WECHSEL
angezeigt wird, um den Modus zu deaktivieren.
Der Wert wechselt auf „AUS“ in blauer Farbe.

, während das Symbol
Drücken Sie
LÖSCHEN angezeigt wird, um den Wechsel derEmpfindlichkeitseinstellung abzubrechen.

autoSAT Modus Aus Ausgewählt

Drücken Sie, während das Symbol
SPEICHERN angezeigt wird, um das Ausschalten
des autoSAT-Modus zu bestätigen. Der Wert
wechselt auf „AUS“ in schwarzer Farbe.
Empfindlichkeits-Modus Einstellungswert 3 Bestätigt

, während das Symbol
Drücken Sie
LÖSCHEN angezeigt wird, um das Ausschalten
des autoSAT-Modus abzubrechen.

Empfindlichkeits-Modus Einstellungswert 5 (max. Wert) Bestätigt

Indem Siedrücken, wenn auf dem Bildschirmdas Symbol „Weiter“ angezeigt wird, wechselt der
Bildschirm zur nächsten Menü-Option.

autoSAT-Modus Aus Bestätigt

Wiederholen Sie diese Vorgänge und wechseln Sie
auf EIN, um den autoSAT-Modus einzuschalten.
Indem Siedrücken, wenn auf dem Bildschirmdas Symbol „Weiter“ angezeigt wird, wechselt der
Bildschirm zur nächsten Menü-Option.
PN MN232-C4 L | Bedienungsanleitung DE- 95

FreeStyle Comfort
Fehlerbehebung
ProblemMögliche Ursache

LösungDer FreeStyle Comfort
zeigt die Warnung an und
gibt einen intermittierenden
Signalton aus.

Siehe Warnzustände-Tabelle.

Siehe Warnzustände-Tabelle.

Der Sauerstoffkonzentrator
lässt sich nicht einschalten.

Der Akku ist vollständig
entladen.

Das Gerät über die Netzsteckdose mit Stromversorgen.

Nicht an externer
Stromversorgung
angeschlossen.

Sicherstellen, dass die externen Anschlüsse
sicher verbunden sind.

Allgemeine Funktionsstörung.

Führen Sie einen vollständigen Leistungszyklus
durch, indem Sie den Akku entfernen
und die externe Wechselstrom- oder
Gleichstromversorgung trennen. Setzen Sie
den Akku anschließend wieder ein und derLeistungszyklus ist abgeschlossen. Schließen
Sie bei Bedarf die externe Stromversorgung
wieder an.
Kontaktieren Sie Ihren Geräteanbieter undsteigen Sie gegebenenfalls auf eine andere
Sauerstoffquelle um.

Alle übrigen Probleme.

Führen Sie einen vollständigen Leistungszyklus
durch, indem Sie den Akku entfernen
und die externe Wechselstrom- oder
Gleichstromversorgung trennen. Setzen Sie
den Akku anschließend wieder ein und derLeistungszyklus ist abgeschlossen. Schließen
Sie bei Bedarf die externe Stromversorgung
wieder an.
Steigen Sie auf eine andere Sauerstoffquelle um
und kontaktieren Sie Ihren Geräteanbieter.

Hinweis: Einige Warnungen führen dazu, dass das Gerät heruntergefahren wird.
Hinweis: Während eines totalen Stromausfalls wird das Warnsystem zurückgesetzt (sämtliche vorherigen
Warnungen werden gelöscht) und es wird nur dann eine Warnung angezeigt, wenn der Zustand nach derWiederherstellung der Stromversorgung noch immer besteht.

96 - DE PN MN232-C4 L | Bedienungsanleitung

FreeStyle Comfort
Reiseinformationen
Der FreeStyle Comfort erfüllt die Richtlinien der
FAA (Bundesluftfahrtbehörde der Vereinigten
Staaten) zur Verwendung in Passagierflugzeugen.
Im Vorfeld einer jeden Reise (mit dem Flugzeug,dem Zug, Fähre usw.):• Vergewissern Sie sich, dass sich der Konzentrator
in einem betriebsbereiten Zustand befindet.
• Setzen Sie sich mindestens 48 Stunden vor
dem Abflug mit dem Transportunternehmen in
Verbindung und teilen Sie diesem mit, dass Sie
mit einem FreeStyle Comfort reisen werden.

1. Schalten Sie den POC unter Verwendung des
Akkus ein, wenn keine externe Stromquelleangeschlossen ist.

2. Aktivieren Sie den Flugzeug-Modus nach einer
der folgenden Methode:Es wird empfohlen, auf Reisen stets ReserveAkkupacks mitzuführen. Achten Sie bei Flugreisen
darauf, dass Sie über genügend Akkukapazität fürBodenzeiten (Check-In, Security usw.) und eventuelle
Zwischenlandungen verfügen. Bei den meisten
Fluggesellschaften ist es Pflicht, eine ausreichende
Zahl an vollständig geladenen Akkupacks zur
Stromversorgung des Geräts mitzuführen, welche
für 150 % der erwarteten maximalen Reisezeit(Flugdauer, gesamte Bodenzeit, d. h. Zeiten vor undnach dem Flug sowie während des Umsteigens undunerwartete Verspätungen) ausreicht.

a. Folgen Sie den Anweisungen zum Aktivieren
des Flugzeug-Modus, die Sie unter dem
entsprechenden Abschnitt der Anweisungen zur
Menüführung finden.

Hinweis: Beim Fliegen empfiehlt es sich, die ReserveAkkupacks in einer sicheren Tragetasche mitzuführen.

b. Halten Sie die Aufwärts- und Abwärtstaste
gleichzeitig 5 Sekunden lang gedrückt, bis ein
langer Signalton zu hören ist und das FlugzeugSymbol angezeigt wird.

BESTIMMUNGEN DER
US-LUFTFAHRTBEHÖRDEErsatzakkus müssen einzeln gegen Kurzschlüsse
geschützt werden. Hierzu sind sie entweder in
die ursprüngliche Verkaufsverpackung zu legenoder die Kontaktflächen sind anderweitig zu
isolieren, z. B. durch ein Abkleben mit Klebeband,oder die Unterbringung der Akkus in separaten
Plastiktüte oder Schutztasche. Akkus dürfen nur im
Handgepäck mitgeführt werden.

FLUGZEUG-MODUS AKTIVIEREN

3. Wenn das Flugzeug-Symbol angezeigt wird,
stecken Sie den POC ein.

Akkubetrieb
Ein POC kann in einem Flugzeug gemäß den
normalen Betriebsverfahren ausschließlich im
Akkubetrieb verwendet werden, wenn keine
externe Stromquelle angeschlossen ist.
Externe Stromversorgung
Wenn in einem Flugzeug eine Steckdose zur
Verfügung steht, die betriebsbereit ist und deren
Standort für die Verwendung als POC-Stromquellegeeignet ist, kann sie als Ersatz für POC-Akkus
dienen. In diesem Fall ist das folgende empfohlene
Verfahren für den Übergang von Akku- zu
Flugzeugstrom zu befolgen:

Hinweis: Aufgrund der Beschränkungen in Flugzeugen
laden sich die Akkus im Flugzeug-Modus nicht auf.

PN MN232-C4 L | Bedienungsanleitung DE- 97

FreeStyle Comfort
Zubehör

VERSTAUEN DES GERÄTS IN DER
TRAGETASCHE

Um einen ordnungsgemäßen und sicheren Betrieb
zu gewährleisten, bitte ausschließlich folgende
Zubehörteile von CAIRE verwenden, die Ihnen vonIhrem Sauerstoff-Versorger bereitgestellt werden.
Die Verwendung von Zubehörteilen, die hier nicht
aufgeführt sind, kann die Funktion und/oder die
Sicherheit des Geräts nachhaltig beeinträchtigen.

1. Stecken Sie das Gerät in den Boden der Tasche.
Achten Sie dabei darauf, dass die Öffnungen fürFilter und Akku korrekt zueinander ausgerichtet
sind.

FreeStyle Comfort Standard Zubehör
Tragetasche

MI396-1

Rucksack

MI459-1

Schultergurt

MI397-1

Wechselstrom-Netzteil (sämtliche
Kabel enthalten)

PW036-1S

Gleichstrom-Netzkabel

CD041-1

Externer Grobpartikelfilter (rechts)

FI226-1

Externer Grobpartikelfilter (links)

FI227-1

2. Schließen und sichern Sie den Reißverschluss
am unteren Teil der Tasche.

Verfügbares zusätzliches Zubehör
Zusätzlicher 8-Zellen Akkupack

BT034-1S

16-Zellen-Akkupack

BT035-1S

Tischladegerät

BT036-1

Zubehörtragetasche

MI372-2

Hinweis: Landesspezifische Stecker sind, sofern
oben angegeben, ebenfalls erhältlich. Falls weitere
Ersatzteile bestellt werden müssen, wenden Sie sichbitte an CAIRE oder an Ihren Geräteanbieter.

Bedienungsanleitung Tragetasche
Um Ihren FreeStyle Comfort in der Tragetasche
zu verstauen und zu sichern, befolgen Sie die
folgenden Anweisungen.

98 - DE PN MN232-C4 L | Bedienungsanleitung

ANBRINGEN DES TRAGEGURTES
1. Der Schultergurt kann entweder direkt am Gerätoder an der Tragetasche befestigt werden.
2. Befestigen Sie den Gurt am Ring des Geräts oderder Tasche.

FreeStyle Comfort
Entsorgung und Wiederverwendung Materialien mit Direktkontaktzum Benutzer.
Geben Sie FreeStyle Comfort, einschließlich aller
Komponenten, immer zur ordnungsgemäßen
Entsorgung an Ihren Geräteanbieter zurück. FürAnweisungen zur ordnungsgemäßen Entsorgung derBatterie können Sie sich zudem an die zuständigen
Stellen Ihres Wohnortes wenden. Das Gerät kannzwischen Personen wiederverwendet werden,muss jedoch zur Wiederaufbereitung an den
Gerätehersteller zurückgegeben werden.

WEEE und RoHS
Das Symbol soll den Besitzer des Geräts
darauf hinweisen, dass das Gerät gemäß der Richtlinie über Elektro- und Elektronik-Altgeräte amEnde seiner Lebensdauer zu einer Recyclingstelle
gebracht werden muss.

• Konzentratorgehäuse: PC/ABS• Bedienfeld des Konzentrators: Polyester EBG7
oder gleichwertig• Sauerstoff-Auslass, Düse: Aluminium• Geräteetikett: Lexan• Kabelanschlüsse: Polykarbonat/Vinylchlorid• Netzkabel: PVC, Metall• Stromversorgung: Lexan 940 (Polycarbonat)• Akku-Pack: PC/ABS• Netzteil- und Akkupack-Etiketten: Polyesterfolie• Tragetasche für den Konzentrator: Cordura,
Neopren, Nylonfutter• Tragegurt: Polypropylen, Acetyl, Vinyl, Cordura,
NeoprenUnsere Produkte entsprechen der Richtlinie zur Beschränkung der Verwendung bestimmter gefährlicher Stoffe (RoHS). Sie enthalten Blei oder andere
gefährliche Materialien höchstens in Spuren.
VORSICHT: Hinweise zur ordnungsgemäßen Entsorgung Ihres Akkus erhalten
Sie von dem Bereitsteller, der Ihnen das
Gerät zur Verfügung gestellt hat, undvon den für die Entsorgung von Abfällen
zuständigen Ämtern.
VORSICHT: Das Molekularsieb immer
gemäß allen Bundes-, Landes- und lokalen
Umweltkontrollvorschriften entsorgen.

PN MN232-C4 L | Bedienungsanleitung DE- 99

FreeStyle Comfort
EMV-Prüfung
Leitlinien und Herstellererklärung – elektromagnetische Aussendungen
Der FreeStyle Comfort ist für den Gebrauch in einer elektromagnetischen Umgebung bestimmt, wie sie
nachfolgend beschrieben ist. Der Kunde oder der Benutzer des FreeStyle Comfort sollte sicherstellen,
dass das Gerät in einer entsprechenden Umgebung verwendet wird.
StöraussendungsmesÜbereinstimmung Elektromagnetische Umgebung – Leitlinien
sungenDer FreeStyle Comfort nutzt HF-Energie nur für interne
HF-Aussendung
Funktionen. Daher ist die HF-Aussendung sehr gering, und es ist
Gruppe 1
unwahrscheinlich, dass benachbarte elektronische Geräte gestört
CISPR 11werden.
HF-Aussendung
Klasse B
CISPR 11
Der FreeStyle Comfort ist für den Gebrauch in allen Einrichtungen
Aussendungen voneinschließlich denen im Wohnbereich und solchen geeignet,
Oberschwingungen
Klasse A
die unmittelbar an ein öffentliches Niederspannungsnetz
nach IEC 61000-3-2
angeschlossen sind, das auch Gebäude versorgt, die zu
Aussendungen von
Wohnzwecken benutzt werden.
Spannungsschwanstimmt überein
kungen/Flicker nachIEC 61000-3-3

100 - DE PN MN232-C4 L | Bedienungsanleitung

FreeStyle Comfort
Leitlinien und Herstellererklärung – elektromagnetische Störfestigkeit

Der FreeStyle Comfort ist für die Verwendung in der unten angegebenen elektromagnetischen Umgebungvorgesehen. Der Kunde oder der Benutzer des FreeStyle Comfort muss sicherstellen, dass er in einer solchen
Umgebung verwendet wird.
Störfestigkeitsprü- IEC 60601 PrüfpegelÜbereinstimmungsElektromagnetische
fung
pegel
Umgebung – Leitlinien
Fußböden sollten aus Holz oder Beton
Elektromagnetischebestehen oder mit Keramikfliesen
Umgebung –±8 kV Kontakt±8 kV Kontakt
versehen sein. Wenn der Fußboden mit
Leitlinien
synthetischem Material versehen ist,
±15 kV Luftaustritt±15 kV Luftaustrittmuss die relative Luftfeuchte mindestens
IEC 61000-4-2
30 % betragen.
Schnelle transiente ±2 kV für Netzleitungen ±2 kV für Netzleitungen Die Qualität des Stromnetzes sollte
der eines typischen professionellen
elektrische StörgröGesundheitszentrums und derßen/Bursts±1 kV für Eingangs- undhäuslichen Gesundheitsversorgung
N/A
IEC 61000-4-4
Ausgangsleitungenentsprechen.
±1 kV Gegentaktspannung

Stoßspannungen
IEC 61000-4-5±2 kV Gleichtakt auf
Wechselstromleitungen±1 kV Differential auf
Wechselstromleitungen±1 kV Gegentaktspannung

Die Qualität des Stromnetzes sollte
der eines typischen professionellen
Gesundheitszentrums und derhäuslichen Gesundheitsversorgung
entsprechen.

±2 kV Gleichtakt
auf I/O-Leitungen im
Außenbereich
0 % UT für 0,5 Zyklen
0 % UT für 0,5 Zyklen
Die Qualität des Stromnetzes sollte(0°, 45°, 90°, 135°,(0°, 45°, 90°, 135°,
180°, 225°, 270°, 315°) 180°, 225°, 270°, 315°) der eines typischen professionellen
Gesundheitszentrums und derSpannungseinbrühäuslichen Gesundheitsversorgung
che, Kurzzeitunter0 % UT für 1 Zyklus (0°) 0 % UT für 1 Zyklus (0°) entsprechen. Wenn der Benutzer desbrechungen und
FreeStyle Comfort den Betrieb bei
Schwankungender Versorgungs70 % UT (30 % Dip in
70 % UT (30 % Dip in UT) Stromausfällen fortsetzen muss, wird
empfohlen, den FreeStyle Comfort
spannung gemäß
UT) für 25/30 Zyklen (0°) für 25/30 Zyklen (0°)über eine unterbrechungsfreie
IEC 61000-4-11
Stromversorgung oder einen Akku
0 % UT für 250/300
0 % UT für 250/300
zu betreiben.
Zyklen (0°)
Zyklen (0°)
Die Magnetfelder der Netzfrequenz
Magnetfeld bei
sollten auf einem Niveau liegen, dasder Netzfrequenzcharakteristisch für einen gewöhnlichen
3 A/m
3 A/m
(50/60 Hz) gemäßStandort in einer professionellen
IEC 61000-4-8
Gesundheitseinrichtung und im
häuslichen Gesundheitswesen ist.
HINWEIS: UT ist die Netzwechselspannung vor der Anwendung des Testpegels.

HINWEIS: In einigen Fällen kann es vorkommen, dass der FreeStyle Comfort nach einer signifikanten
elektrostatischen Entladung zurückgesetzt wird und ein vollständiger Leistungszyklus erforderlich ist, um den
normalen Betrieb wieder aufzunehmen. Weitere Informationen hierzu finden Sie im Abschnitt Fehlerbehebung.

PN MN232-C4 L | Bedienungsanleitung DE- 101

FreeStyle Comfort
Leitlinien und Herstellererklärung – elektromagnetische Störfestigkeit

Der FreeStyle Comfort ist für die Verwendung in der unten angegebenen elektromagnetischen Umgebung vorgesehen. Der Kunde oder der Benutzer des FreeStyle Comfort muss sicherstellen, dass er in einer solchen Umgebung
verwendet wird.
Störfestigkeits- IEC 60601 PrüfpegelÜbereinstimmungsElektromagnetische
prüfungpegel
Umgebung – Leitlinien
10 Vrms
Tragbare und mobile HF10 Vrms
Geleitete HF
Kommunikationsgeräte sollten
150 kHz bis 80 MHz
IEC 61000-4-6
nicht näher an irgendeinem Teil des
FreeStyle Comfort, einschließlich
der Kabel, verwendet werden als derempfohlene Abstand, der sich aus der
10 V/m
10 V/mGestrahlte HF
für die Frequenz des Senders geltenden
80 MHz bis 2700 MHz
IEC 61000-4-3
Gleichung ergibt.
Empfohlener Schutzabstand= 1.2 Pd
= 1.2 P = 80 MHz bis 800 MHzdd= 1.2 P = 800 MHz bis 2700 MHz
wobei P die maximale Ausgangsleistungdes Senders in Watt (W) nach Angaben
des Senderherstellers und d derempfohlene Abstand in Metern (m) ist.
Die Feldstärken von ortsfesten
HF-Sendern, die durcheine elektromagnetischeStandortuntersuchung ermittelt wurden, asollten in jedem Frequenzbereich unterdem Konformitätsniveau liegen.b
In der Umgebung von Geräten, diedas folgende Bildzeichen tragen, sind
Störungen möglich:

HINWEIS 1: Bei 80 MHz und 800 MHz gilt der höhere Frequenzbereich.
HINWEIS 2: Diese Leitlinien sind möglicherweise nicht in allen Fällen anwendbar. Die Ausbreitungelektromagnetischer Wellen wird durch Absorptionen und Reflexionen von Gebäuden, Gegenständen
und Menschen beeinflusst.
aFeldstärken von ortsfesten Sendern, wie Basisstationen für Funktelefone (Mobil-/schnurlose Telefone) und mobile
Landfunkgeräte, Amateurfunk, AM- und FM-Rundfunk und Fernsehsendungen können theoretisch nicht genau
vorhergesagt werden. Um die elektromagnetische Umgebung hinsichtlich der stationären Sender zu ermitteln, sollte
eine Standortanalyse erwogen werden. Wenn die gemessene Feldstärke an dem Standort, an dem der FreeStyle
Comfort benutzt wird, die obigen Übereinstimmungspegel überschreitet, sollte der FreeStyle Comfort beobachtet
werden, um die bestimmungsgemäße Funktion zu überprüfen. Wenn ungewöhnliche Leistungsmerkmale
beobachtet werden, können zusätzliche Maßnahmen erforderlich sein, z. B. eine Neuausrichtung oder ein anderer
Standort des FreeStyle Comfort.
b

Im Frequenzbereich von 150 kHz bis 80 MHz muss die Feldstärke weniger als 10 V/m betragen.

102 - DE PN MN232-C4 L | Bedienungsanleitung

FreeStyle Comfort
Empfohlene Schutzabstände zwischen tragbaren und mobilen
HF-Kommunikationsausrüstungen und dem FreeStyle Comfort

Der FreeStyle Comfort ist für die Verwendung in einer elektromagnetischen Umgebung vorgesehen, in derabgestrahlte HF-Störungen kontrolliert werden. Der Kunde oder der Benutzer des FreeStyle Comfort kann dazu
beitragen, elektromagnetische Störungen zu vermeiden, indem er einen Mindestabstand zwischen tragbaren
und mobilen HF-Kommunikationsgeräten (Sendern) und dem FreeStyle Comfort einhält, wie unten empfohlen,
entsprechend der maximalen Ausgangsleistung der Kommunikationsgeräte.
Nennleistung desTrennungsabstand je nach Frequenz des Senders
Sendersm
150 kHz bis 80 MHz

80 MHz bis 800 MHz

800 MHz bis 2,5 GHzW

d = 1.2 P
d = 1.2 P
d = 1.2 P
0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
Für Sender, deren maximale Nennleistung in obiger Tabelle nicht angegeben ist, kann der empfohlene Schutzabstand (d)in Metern (m) unter Verwendung der Gleichung ermittelt werden, die für die jeweilige Sendefrequenz gilt. Dabei ist P die
maximale Nennleistung des Senders in Watt (W) gemäß Angabe des Senderherstellers.
HINWEIS 1: Bei 80 MHz und 800 MHz gilt der Schutzabstand für den höheren Frequenzbereich.
HINWEIS 2: Diese Leitlinien sind möglicherweise nicht in allen Fällen anwendbar. Die Ausbreitung elektromagnetischer
Wellen wird durch Absorptionen und Reflexionen von Gebäuden, Gegenständen und Menschen beeinflusst.

Klassifikation

Art des Schutzes vor elektrischem Schlag:
Klasse II	Schutz vor Stromschlag ist			erreicht durch doppelte Isolierung.
Grad des Schutzes vor elektrischem Schlag:	Typ BF Ausrüstung, die einen bestimmten
		Grad des Schutzes vor		Stromschlägen bietet bezüglich		1) zulässigem Verluststrom;		2) Zuverlässigkeit der Schutzleiterverbindung		(falls vorhanden).
		Nicht für die Direktanwendung am Herzen
geeignet.
Unabhängige Prüfung gemäß den Normen
für medizinische elektrische Geräte:	Getestet auf Konformität mit• IEC 60601-1 Auflage 3.1: Medizinischeelektrische Geräte – Teil 1: Allgemeine
Anforderungen an die Sicherheit.
• CAN/CSA C22.2 Nr. 60601-1-8 Auflage 2.1,
Medizinische elektrische Geräte – Teil 1:
Allgemeine Sicherheitsanforderungen
Schutz gegen potenzielle elektromagnetischeoder andere Störungen zwischen den Geräten
und anderen Geräten.

• Geprüft auf Konformität mit EN 60601-1-2 /IEC 60601-1-2: 4. Auflage.
	• Getestet auf Konformität mit RTCA/DO160
	Abschnitt 21 Kategorie M.
• CISPR 11 / EN 55011 Klasse B Gruppe 1,	„Industrie, Wissenschaft und Medizin (ISM)	Geräte“• FCC Teil 15, Unterabschnitt B – Klasse B	Unbeabsichtigte Strahler
IP22 - Schutz gegen das Eindringen von Fingern
oder ähnlichen Gegenständen mit einem Durchmesser von mehr als 12,5 mm. Vertikal tropfendes
Wasser darf keine schädliche Wirkung haben,
wenn das Gehäuse in einem Winkel von 15 Gradvon seiner normalen Position geneigt ist.
Zulässige Reinigungs- und Infektionskontrollmethoden: Siehe Abschnitt „Reinigung, Pflege undordnungsgemäße Wartung“ in dieser FreeStyle-Gebrauchsanleitung.
Grad der Anwendungssicherheit bei Vorhandensein
entzündlicher anästhetischer Gase:	Die Geräte sind für solche Anwendungen nicht
geeignet.
Betriebsmodus: Dauerbetrieb.
PN MN232-C4 L | Bedienungsanleitung DE- 103

FreeStyle Comfort
Controles del usuario e indicadores de estado del sistema
ISO 7000
Manual del operador, instrucciones defuncionamiento. Reg. n.º 1641

ISO 7010: Símbolos gráficos: colores y símbolos deseguridad; símbolos de seguridad registrados
Debe leer el manual de instrucciones.
Reg. n.º M002

País y fecha de fabricación. El «CC»
identifica el código de país de dos letrasdel país de fabricación. La fecha defabricación está en formato AAAAMM-DD. Reg. n.º 6049

Mantener alejado de las llamas, delfuego y de las chispas. No acercar afuentes de ignición abiertas. Se prohíbe
fumar. Reg. n.º P003

Mantenerlo alejado de la lluvia.
Mantenerlo seco. Reg. n.º 0626

No fumar cerca de la unidad ni mientras
la esté utilizando. Reg. n.º P002

Límite de apilamiento por número,
donde «n» indica el número máximo deunidades permitidas. Reg. n.º 2403
Número de catálogo. Reg. n.º 2493

Advertencia. Reg. n.º W001
IEC 60417
Parte aplicada tipo BF (grado deprotección frente a descargas eléctricas).
Reg. n.º 5333

Número de serie. Reg. n.º 2498
Intervalo de temperatura defuncionamiento o almacenamiento. Reg.
n.º 0632Intervalo de humedad dealmacenamiento. Reg. n.º 2620
Limitación de presión atmosférica. Reg.
n.º 2621
Nombre y domicilio del fabricante. Reg.
n.º 3082

Equipo de Clase II, Doble aislamiento,
n.º de reg. 5172
Reglamento de la UE 2017/745
Representante autorizado en la
Comunidad Europea
Si la etiqueta del identificador único dedispositivo del producto (UDI) incluye
el símbolo CE ####, el dispositivo
cumple con los requisitos sobre
dispositivos médicos. El símbolo CE#### indica el número de organismo
notificado.
Símbolos adicionales

Este lado hacia arriba. Reg. n.º 0623

Mantener la unidad lejos de materiales
inflamables, aceite y grasa.

Frágil. Manipular con cuidado.
Reg. n.º 0621

No desmontar.

Contiene sustancias peligrosas.
Reg. n.º 3723

Aumentar flujo.

Importador. Reg. n.º 3725

Disminuir flujo.

Salida de O2

Botón Menú.
Interruptor de encendido.
Reg. n.º 5009
Cumple con
RTCA/DO160 Sección
21 CategoríaM.

104 - ES PN MN232-C4 L | Manual del usuario

El fabricante de esta prueba de concepto
ha determinado que este dispositivo
cumple todos los criterios de aceptación
aplicables de la FAA y su uso a bordo deaeronaves.

FreeStyle Comfort
El fabricante de esta prueba de concepto
ha determinado que este dispositivo
cumple todos los criterios de aceptación
aplicables de la FAA y su uso a bordo deaeronaves.

RAEE

Este símbolo pretende recordar a lospropietarios de los equipos que deben
llevarlos a un centro de reciclado al
final de su vida útil, de conformidad
con la Directiva sobre residuos deaparatos eléctricos y electrónicos(RAEE). Nuestros productos
cumplirán con la Restricción sobre
Sustancias Peligrosas (RoHS).
No contendrán más que trazasde plomo o de otros materiales
peligrosos.

Símbolo de reciclaje
Representante autorizado en Suiza(ADD)
Si el dispositivo lleva la marca UKCA,
como se muestra con UKCA####que indica el número del organismo
notificado, este dispositivo cumple conla normativa UKCA.
21 CFR 801.15
La legislación federal limita la venta deeste dispositivo a médicos o por orden
médica.

IEC 60601-1-8
Alarma técnica de baja prioridad
ISO 15223-1

Certificación ETL
Certificado para los mercados de
EE. UU. y Canadá, según las normas
vigentes en ambos países
Contiene la ID de la FCC: WAP2001
Aviso de la FCC
Este dispositivo puede contener CYBLE-022001-00,
incluida la antena 2450AT18B100 de Johnson
Technology, cumple con la parte 15 de las reglas dela FCC. El dispositivo cumple los requisitos para laaprobación del transmisor modular, tal y como se
detalla en el aviso público de la FCC DA00-1407. El
funcionamiento del transmisor está sujeto a las dos
condiciones siguientes: (1) El dispositivo no debe
causar interferencias dañinas y (2) Este dispositivo
debe aceptar cualquier interferencia recibida,incluidas las interferencias que puedan causar un
funcionamiento no deseado. El producto puede
estar cubierto por una o más patentes, de EE. UU. ointernacionales. Visite nuestro sitio web para obtenerel listado de patentes aplicables.
Pat.: www.caireinc.com/corporate/patents

Dispositivo médico. Reg. n.º 5.7.7
Identificador único del dispositivo
n.º 5.7.10
A S/NZS 4417.1, AS/NZS 4417.2
Marca de cumplimiento normativode ACMA: indica el cumplimientode la normativa de ACMA (ADD)

Tecnología inalámbrica
Tipos de conexión

SSP, iAP2, GATT

Frecuencia

De 2402 a 2480 MHz

Salida de potencia deradiofrecuencia máxima

+4 dBmRango de funcionamiento

10 m (clase 2)

Se recomienda que la máquina se halle a una distancia mínima de 1,1 cm (0,43 pulg.) del cuerpo durante
el funcionamiento. Esto no procede con máscaras,
tubos de aire ni accesorios.

Símbolos gráficos a utilizar en el equipo IEC 60417
Corriente continua, n.º de reg. 5031
Corriente alterna, n.º de reg. 5032
IEC 60601-1
A prueba de goteos.
Directiva del Consejo 2012/19/UE

PN MN232-C4 L | Manual del usuario ES - 105

FreeStyle Comfort
Desembalar su FreeStyle® Comfort™
Compruebe que todos loscomponentes enumerados están
incluidos en el paquete. Si falta
algún elemento, póngase en
contacto con el proveedor desu concentrador de oxígeno
inmediatamente.

Accesorios estándares• Concentrador de oxígeno portátil• Bolso de transporte• Correa para hombro• Paquete de batería de 8 celdas• Fuente de alimentaciónde CA o CC
• Cable de alimentación de CA• Fuente de alimentación de CC• Manual del usuario

Accesorios opcionales• Paquete de batería de 8 celdas
adicional• Batería de 16 celdas• Cargador de escritorio• Bolsa para llevar todos losaccesorios

Conocer su FreeStyle Comfort
Familiarícese con las nuevas funciones clave del FreeStyle Comfort y su interfaz de usuario.

Vista frontal
Fijación de la correa

Interfaz de usuario

Salida deoxígenoSalida de aire

Entrada de aire/
Filtros
Entrada de aire/
Filtros
Pestaña delreceptáculo
del puerto dealimentación
Batería
Botón de extracciónde la batería
106 - ES PN MN232-C4 L | Manual del usuario

FreeStyle Comfort
Vista superior
Pantalla
Aumentar flujo

Botón deencendido

Botón Menú

Disminuir flujo

Pantalla con información de la batería
Ajuste de flujo

Indicadorde batería

Indicador autoSAT

Pantalla de alarma

Porcentaje debatería restante

Indicador desuministro bolo

Pantalla con información de alimentación CA/CC

Ajuste de flujo
Indicadorde batería

Espacio delindicador dealarmaIndicador de carga

PN MN232-C4 L | Manual del usuario ES - 107

FreeStyle Comfort
Qué es el concentrador de oxígeno
El aire que respiramos contiene aproximadamente
un 21 % de oxígeno, un 78 % de nitrógeno y un 1 %de otros gases. En el FreeStyle® Comfort®, el aireambiente entra a la máquina a través de las entradasde aire. A continuación, pasa a través de un material
absorbente llamado tamiz molecular. Este material
separa el oxígeno del nitrógeno y permite que solo
el oxígeno pase a través de él. El resultado es elsuministro al usuario de un flujo de oxígeno de altaconcentración.
El FreeStyle Comfort combina tecnologías avanzadas
de concentración de oxígeno y de conservación deoxígeno para crear un dispositivo portátil que permite
la mayor portabilidad y movilidad. La administraciónavanzada de flujo de pulso detecta rápidamente cuando
inspira el usuario y administra un pulso de oxígeno
al principio de cada inhalación. La administraciónde flujo de pulso es idónea para realizar actividades
y pasar tiempo fuera de casa. Conserva el oxígeno y
alarga la duración de la batería, a diferencia del flujo
continuo. El FreeStyle Comfort es tan ligero quepuede transportarse mediante la correa para hombro.
Puede funcionar con alimentación de CA (pared), CC(vehículo) o de batería.

Uso previsto
El concentrador de oxígeno CAIRE FeeStyle Comfort
está indicado para la administración de oxígeno
suplementario. El dispositivo no está destinado parasu uso como soporte vital ni proporciona capacidad desupervisión del paciente.

Por qué su médico le ha prescrito
oxígenoMuchas personas padecen multitud de enfermedades
cardíacas, pulmonares y otras enfermedades
respiratorias. Un número considerable de estas
personas puede beneficiarse del tratamiento con
oxígeno suplementario en casa, al viajar o mientras
realiza actividades cotidianas fuera de casa.
El oxígeno es un gas que constituye el 21 % del aireambiente que respiramos, y el cuerpo humano dependedel suministro constante de este para funcionar
correctamente. Su médico le ha prescrito un flujo o
una configuración para tratar su trastorno respiratorio
particular.

108 - ES PN MN232-C4 L | Manual del usuario

Aunque el oxígeno no se trata de una droga adictiva, laoxigenoterapia sin autorización puede ser peligrosa.
Debe consultar con un médico antes de utilizar elconcentrador de oxígeno. El proveedor de equipos que
suministra el equipo de oxígeno le mostrará cómo ajustar
el caudal que se le ha prescrito.

Perfil del operador
Los concentradores están destinados a suministrar
oxígeno suplementario a los usuarios con molestias
debidas a alteraciones que afectan a la eficiencia de sus
pulmones para transferir el oxígeno en el aire a su torrente
sanguíneo. Los concentradores de oxígeno móviles(POC) no almacenan ni contienen oxígeno. No necesitan
ser rellenados y pueden recargarse en cualquier fuente dealimentación de CA o CC disponible (salvo en un avión).
Así, el usuario puede ser relativamente autosuficiente
en términos de uso doméstico, en movimiento (tanto
dentro como fuera de casa), movilidad y estilo de vida en
general. El uso del concentrador de oxígeno requiere una
prescripción médica.
El usuario habitual de la oxigenoterapia es mayor de
65 años y padece diversas enfermedades respiratorias,
incluida la enfermedad pulmonar obstructiva crónica
(EPOC). Por lo general, los usuarios presentan buenas
capacidades cognitivas, y deben ser capaces decomunicar su incomodidad. Si el usuario es incapazde comunicar su incomodidad, o es incapaz de leer
y comprender el etiquetado y las instrucciones de
uso del concentrador, el uso se recomienda solo bajo
la supervisión de alguien que pueda. Si se siente
alguna molestia durante el uso del concentrador,
se recomienda a los usuarios que se pongan en contacto
con su proveedor de atención médica. Asimismo,se recomienda a los usuarios que dispongan de una
fuente de oxígeno de respaldo (por ejemplo, un tanquede oxígeno) para contar con un suministro de oxígeno
en caso de que se produzca un corte eléctrico o una
avería del concentrador. El uso del concentrador deoxígeno no requiere de otras habilidades o capacidades
singulares del usuario.

FreeStyle Comfort¡Importante!
Las instrucciones de seguridad se definen de lasiguiente manera:
ADVERTENCIA: INFORMACIÓN
DE SEGURIDAD IMPORTANTE SOBREPELIGROS QUE PUEDEN CAUSARLESIONES GRAVES.
PRECAUCIÓN: Información importantesobre la prevención de daños en elsistema FreeStyle Comfort.
Aclaración: Información que requiere atención especial.

Indicaciones de uso
ADVERTENCIA: ES MUY IMPORTANTESELECCIONAR SOLO EL NIVEL DEOXÍGENO PRESCRITO. NO CAMBIE LA
SELECCIÓN DE FLUJO A MENOS QUE UN
PROFESIONAL MÉDICO ACREDITADO ASÍ
SE LO HAYA INDICADO.
ADVERTENCIA: ESTE DISPOSITIVO NO SE HAPROBADO NI AUTORIZADO PARA SU USO CONPACIENTES PEDIÁTRICOS.
ADVERTENCIA: LAS LEYES FEDERALES (DE LOSESTADOS UNIDOS) LIMITAN LA VENTA O ALQUILER
DE ESTE DISPOSITIVO POR PARTE DE UN MÉDICOU OTRO PROVEEDOR DE ATENCIÓN MÉDICA
ACREDITADO.
ADVERTENCIA: LA UNIDAD NO DEBE USARSECOMO SOPORTE VITAL. PUEDE SER NECESARIAUNA SUPERVISIÓN ADICIONAL PARA CUALQUIERUSUARIO QUE SEA INCAPAZ DE COMUNICAR SUINCOMODIDAD MIENTRAS USA ESTE DISPOSITIVO.
LOS USUARIOS CON TRASTORNOS AUDITIVOS O
VISUALES PUEDEN NECESITAR ASISTENCIA PARACONTROLAR LAS ALARMAS.
ADVERTENCIA: LOS AJUSTES DEL MODO DE
DOSIS PULSADA DEBEN DETERMINARSE PARACADA USUARIO INDIVIDUALMENTE, SEGÚN
SUS NECESIDADES EN ESTADO DE REPOSO,
AL REALIZAR EJERCICIO Y AL VIAJAR.
ADVERTENCIA: ESTE DISPOSITIVO ESTÁ DISEÑADO
PARA SU USO EN LA ASISTENCIA SANITARIADOMICILIARIA Y EN ENTORNOS AMBULATORIOSSIMILARES.

ADVERTENCIA: ES POSIBLE QUE ALGUNOSESFUERZOS RESPIRATORIOS DEL PACIENTENO ACTIVEN LA FUNCIÓN DE CONSERVACIÓN.
ADVERTENCIA: EL FREESTYLE COMFORT NO
ESTÁ DESTINADO PARA SU USO EN PACIENTESLARINGECTOMIZADOS.
ADVERTENCIA: LAS MUJERES EMBARAZADASO LACTANTES NO DEBEN USAR LOS ACCESORIOSRECOMENDADOS EN ESTE MANUAL, YA QUEPUEDEN CONTENER FTALATOS.

Contraindicaciones de uso
ADVERTENCIA: EN ALGUNASCIRCUNSTANCIAS, EL USO DE OXÍGENO
NO PRESCRITO PUEDE SER PELIGROSO.
ESTE DISPOSITIVO SOLO DEBERÁ
UTILIZARSE CUANDO LO PRESCRIBA
UN MÉDICO.
ADVERTENCIA: ESTE DISPOSITIVO NO ESTÁDISEÑADO PARA SU USO EN ENTORNOS DERESONANCIA MAGNÉTICA O CERCA DE ELLOS,
COMO MÁQUINAS DE RM.
ADVERTENCIA: NO UTILIZAR EN PRESENCIA DE
ANESTÉSICOS INFLAMABLES.
ADVERTENCIA: AL IGUAL QUE CON CUALQUIERDISPOSITIVO ELÉCTRICO, EL USUARIO PUEDE
EXPERIMENTAR PERÍODOS DE FALTA DEFUNCIONAMIENTO COMO RESULTADO DE
UN CORTE EN LA ENERGÍA ELÉCTRICA O LA
NECESIDAD DE TAREAS DE INSPECCIÓN EN EL
CONCENTRADOR DE OXÍGENO POR PARTE DE
UN TÉCNICO CUALIFICADO. EL CONCENTRADORDE OXÍGENO NO ES APROPIADO PARA USUARIOSQUE EXPERIMENTARÍAN CONSECUENCIASADVERSAS EN LA SALUD COMO RESULTADO DE TALINTERRUPCIÓN TEMPORAL.
ADVERTENCIA: ESTE DISPOSITIVO NO ESTÁAPROBADO PARA SU USO EN TODOS LOSENTORNOS. LA EXPOSICIÓN A DETERMINADAS
FUENTES ELECTROMAGNÉTICAS PUEDE
PROVOCAR INTERFERENCIAS O UN RENDIMIENTOINVOLUNTARIO DEL DISPOSITIVO. EN CASO DE
INTERFERENCIA, EL DISPOSITIVO DEBE APAGARSEINMEDIATAMENTE.

ADVERTENCIA: ESTE DISPOSITIVO ESTÁ DISEÑADO
PARA SER UTILIZADO POR EL PACIENTE QUE
RECIBA OXIGENOTERAPIA.

PN MN232-C4 L | Manual del usuario ES - 109

FreeStyle Comfort
Instrucciones de seguridad
ADVERTENCIA: NO SE PERMITEN
MODIFICACIONES A ESTE EQUIPO.
ADVERTENCIA: ESTE DISPOSITIVO SUMINISTRAOXÍGENO DE ALTA CONCENTRACIÓN, LO QUE LO
HACE MUY INFLAMABLE. NO PERMITA QUE SE
FUME NI LA PRESENCIA DE LLAMAS EXPUESTASDENTRO DE LA MISMA HABITACIÓN EN LA QUE SE
ENCUENTRE (1) ESTE DISPOSITIVO O (2) ALGÚNACCESORIO QUE CONTENGA OXÍGENO. SI NO SE
TIENE EN CUENTA ESTA ADVERTENCIA, PUEDE
PRODUCIRSE UN INCENDIO GRAVE, DAÑOS A LOSBIENES O HASTA LESIONES FÍSICAS O LA MUERTE.
ADVERTENCIA: NO OPERE LA UNIDAD EN UN
ESPACIO RESTRINGIDO O CONFINADO (P. EJ., UNA
CAJA O BOLSO PEQUEÑO) DONDE LA VENTILACIÓN
PUEDE SER LIMITADA. ESTO PUEDE OCASIONARQUE EL CONCENTRADOR DE OXÍGENO SE
SOBRECALIENTE Y QUE SU RENDIMIENTO SE VEAPERJUDICADO.
ADVERTENCIA: EL CONCENTRADOR DEBEUBICARSE DE FORMA TAL QUE SE PUEDA EVITAR
EL HUMO, LOS CONTAMINANTES O LOS GASES
TÓXICOS.
ADVERTENCIA: EN CASO DE QUE OCURRA UN
INCIDENTE GRAVE CON ESTE DISPOSITIVO, EL
USUARIO DEBE INFORMÁRSELO DE INMEDIATO
AL PROVEEDOR O AL FABRICANTE. UN INCIDENTEGRAVE SE DEFINE COMO UNA LESIÓN, LA MUERTEO LA POSIBILIDAD DE CAUSAR UNA LESIÓN O LAMUERTE EN CASO DE QUE SE REPITA EL INCIDENTE.
EL USUARIO TAMBIÉN PUEDE INFORMAR EL
INCIDENTE A LA AUTORIDAD COMPETENTE DEL PAÍSDONDE OCURRIÓ.
ADVERTENCIA: EL USO DE ACCESORIOSDE ADMINISTRACIÓN DE OXÍGENO
NO ESPECIFICADOS PARA ESTECONCENTRADOR DE OXÍGENO PUEDEDISMINUIR SU RENDIMIENTO. LOSACCESORIOS RECOMENDADOS SE
MENCIONAN DENTRO DE ESTE MANUAL.
ADVERTENCIA: SI EL CONCENTRADOR DE OXÍGENO
SE HA CAÍDO, HA SUFRIDO DAÑOS O SE HAEXPUESTO AL AGUA, PÓNGASE EN CONTACTOCON EL PROVEEDOR DEL EQUIPO PARA SUINSPECCIÓN O POSIBLE REPARACIÓN. NO USE EL
CONCENTRADOR DE OXÍGENO SI TIENE EL CABLE
DE ALIMENTACIÓN O EL ENCHUFE DAÑADOS.
ADVERTENCIA: ASEGURE FIRMEMENTE, CON EL
CINTURÓN DE SEGURIDAD O DE OTRA FORMA, EL
CONCENTRADOR DE OXÍGENO EN EL VEHÍCULO
DURANTE SU TRANSPORTE PARA PREVENIR DAÑOSO LESIONES.

110 - ES PN MN232-C4 L | Manual del usuario

ADVERTENCIA: LOS REGLAMENTOS DEL
DEPARTAMENTO DE TRANSPORTE (DOT, POR
SUS SIGLAS EN INGLÉS) ESTADOUNIDENSE Y DE
LAS NACIONES UNIDAS (ONU) REQUIEREN LAEXTRACCIÓN DE LA BATERÍA DEL DISPOSITIVOPARA TODOS LOS VIAJES EN AEROLÍNEAS
INTERNACIONALES CUANDO SE FACTURE EL
CONCENTRADOR DE OXÍGENO COMO EQUIPAJE.
CUANDO SE ENVÍA EL CONCENTRADOR DEOXÍGENO, LA BATERÍA TAMBIÉN DEBE EXTRAERSE
DEL DISPOSITIVO Y EMBALARSE ADECUADAMENTE.
ADVERTENCIA: NO UTILICE ESTE EQUIPO SINANTES LEER Y COMPRENDER ESTE MANUAL. SI NO
COMPRENDE LAS ADVERTENCIAS E INSTRUCCIONES,CONTACTE A SU PROVEEDOR DEL EQUIPO ANTESDE INTENTAR USARLO; DE LO CONTRARIO, PUEDENPRODUCIRSE LESIONES O DAÑOS.
ADVERTENCIA: SI SIENTE MALESTAR O TIENEUNA EMERGENCIA MÉDICA, BUSQUE ASISTENCIAMÉDICA DE INMEDIATO.
ADVERTENCIA: LA OPERACIÓN DELCONCENTRADOR DE OXÍGENO SIN TENEREN CUENTA LAS ESPECIFICACIONES DE LA
TEMPERATURA DE FUNCIONAMIENTO PUEDE
LIMITAR LA CAPACIDAD DEL CONCENTRADOR
DE CUMPLIR CON LAS ESPECIFICACIONES DE
CONCENTRACIÓN DE OXÍGENO. CONSULTE LASECCIÓN DE ESPECIFICACIONES DE ESTE MANUALPARA CONOCER LOS LÍMITES DE TEMPERATURA.
ADVERTENCIA: NO USE ACEITE, GRASA,PRODUCTOS A BASE DE PETRÓLEO NI OTROSPRODUCTOS INFLAMABLES CON LOS ACCESORIOSDE TRANSPORTE DE OXÍGENO O CON EL
CONCENTRADOR DE OXÍGENO. SOLO DEBEN USARSELOCIONES O BÁLSAMOS COMPATIBLES CON ELOXÍGENO, A BASE DE AGUA. EL OXÍGENO ACELERA LACOMBUSTIÓN DE LAS SUSTANCIAS INFLAMABLES.
ADVERTENCIA: EL USO INCORRECTO DE LA
BATERÍA PUEDE OCASIONAR QUE ESTA SE
CALIENTE O SE ENCIENDA, Y PUEDE CAUSARLESIONES SERIAS. ASEGÚRESE DE NO PERFORAR,GOLPEAR, PISAR, DEJAR CAER O SOMETER DEOTRA FORMA LA BATERÍA A IMPACTOS O GOLPESFUERTES. EL USO DE BATERÍAS QUE ESTÉNDAÑADAS PODRÍA CAUSAR LESIONES FÍSICAS.
ADVERTENCIA: NO PROVOQUE UN CORTOCIRCUITO
DE LOS CONTACTOS DE METAL DE LA BATERÍA CONOTROS OBJETOS METÁLICOS, COMO LLAVES O
MONEDAS. ESTO PUEDE HACER QUE SE GENERENCHISPAS O CALOR EXCESIVO.

FreeStyle Comfort
ADVERTENCIA: MIENTRAS USA EL CONCENTRADORDE OXÍGENO PORTÁTIL AL AIRE LIBRE CON LA
FUENTE DE ALIMENTACIÓN DE CA, CONECTE LA
FUENTE DE ALIMENTACIÓN SOLAMENTE A UNA TOMA
CON INTERRUPTOR POR PÉRDIDA A TIERRA (GFI).
ADVERTENCIA: EL CONCENTRADOR DE OXÍGENO
NO DEBE USARSE ADYACENTE A O APILADO CONOTROS EQUIPOS. SI EL USO ADYACENTE O APILADOES INEVITABLE, EL DISPOSITIVO DEBE OBSERVARSEPARA VERIFICAR SU FUNCIONAMIENTO NORMAL.
ADVERTENCIA: EL USO DE CABLES Y ADAPTADORESDIFERENTES A LOS ESPECIFICADOS, CON
EXCEPCIÓN DE LOS CABLES Y DE LOS
ADAPTADORES VENDIDOS POR EL FABRICANTEDEL EQUIPO MÉDICO ELÉCTRICO COMO PIEZASDE REPUESTO PARA COMPONENTES INTERNOS,PUEDE DERIVAR EN UNA MAYOR EMISIÓN DE MENORINMUNIDAD DEL CONCENTRADOR DE OXÍGENO.
ADVERTENCIA: USE ÚNICAMENTE EL VOLTAJE
ELÉCTRICO INDICADO EN LA ETIQUETA DEESPECIFICACIONES ADHERIDA AL DISPOSITIVO.
ADVERTENCIA: NO UTILICE CABLES DE EXTENSIÓN
EN ESTA UNIDAD NI CONECTE DEMASIADOSENCHUFES A LA MISMA TOMA ELÉCTRICA. EL USO
DE CABLES DE EXTENSIÓN PODRÍA AFECTARDE FORMA ADVERSA EL RENDIMIENTO DELDISPOSITIVO. DEMASIADOS ENCHUFES EN UNATOMA PUEDEN OCASIONAR UNA SOBRECARGA DEL
PANEL ELÉCTRICO Y HACER QUE EL DISYUNTOR/FUSIBLE SE ACTIVE O DISPARE SI EL DISYUNTORO EL FUSIBLE NO FUNCIONAN.
PRECAUCIÓN: En caso de alarma o si
observa que el concentrador de oxígeno
no funciona correctamente, consultela sección de resolución de problemasde este manual. Si no puede resolver el
problema, consulte con el proveedor de su
equipo.
PRECAUCIÓN: Para evitar la anulación de la garantía,
siga todas las instrucciones del fabricante.

ADVERTENCIA: NO INTENTE REALIZARTAREAS DE MANTENIMIENTO APARTE
DE LAS SOLUCIONES POSIBLESENUMERADAS EN ESTE MANUAL.
ADVERTENCIA: USE SOLO ACCESORIOSRECOMENDADOS POR EL FABRICANTE. EL USO
DE OTRO ACCESORIO PUEDE SER PELIGROSO,CAUSARÁ DAÑOS GRAVES A SU CONCENTRADORDE OXÍGENO Y ANULARÁ LA GARANTÍA.
ADVERTENCIA: EL FABRICANTE RECOMIENDADISPONER DE UNA FUENTE ALTERNATIVA DEOXÍGENO COMPLEMENTARIO EN CASO DE
CORTE ELÉCTRICO, CONDICIÓN DE ALARMA O
FALLA MECÁNICA. CONSULTE A SU MÉDICO O AL
PROVEEDOR DEL EQUIPO PARA DETERMINAR EL
TIPO DE SISTEMA DE RESERVA QUE NECESITA.
ADVERTENCIA: NO PERMITA QUE SE BLOQUEEN LAENTRADA DE AIRE NI LAS ABERTURAS DE SALIDA
DE AIRE. NO DEJE CAER NI INSERTE OBJETOSEN NINGUNA ABERTURA DEL DISPOSITIVO. ESTO
PUEDE OCASIONAR QUE EL CONCENTRADORDE OXÍGENO SE SOBRECALIENTE Y QUE SU
RENDIMIENTO SE VEA PERJUDICADO.
ADVERTENCIA: FUMAR MIENTRAS SE USA OXÍGENO
ES LA CAUSA PRINCIPAL DE LESIONES POR
INCENDIOS Y MUERTES RELACIONADAS. DEBESEGUIR LAS ADVERTENCIAS DE SEGURIDAD:
ADVERTENCIA: NO PERMITA QUE HAYACIGARRILLOS ENCENDIDOS, VELAS O LLAMASEXPUESTAS EN LA MISMA HABITACIÓN EN QUE SE
ENCUENTRE EL DISPOSITIVO O LOS ACCESORIOSQUE TRANSPORTAN OXÍGENO.
ADVERTENCIA: FUMAR MIENTRAS SE USA UNA
CÁNULA DE OXÍGENO PUEDE CAUSAR QUEMADURAS
FACIALES Y POSIBLEMENTE LA MUERTE.
ADVERTENCIA: SI FUMA, ESTOS 3 PASOS PUEDENSALVAR SU VIDA: APAGUE EL CONCENTRADORDE OXÍGENO, RETIRE LA CÁNULA Y SALGA DE
LA HABITACIÓN DONDE SE ENCUENTRE ELDISPOSITIVO.

PN MN232-C4 L | Manual del usuario ES - 111

FreeStyle Comfort
ADVERTENCIA: DEBEN EXHIBIRSE LETREROS DE«NO FUMAR: OXÍGENO EN USO» EN EL HOGAR,O EN EL SITIO DONDE SE USE EL OXÍGENO. LOS
USUARIOS Y LOS CUIDADORES SANITARIOS DEBEN
RECIBIR INFORMACIÓN ACERCA DE LOS PELIGROS
DE FUMAR AL USAR OXÍGENO MEDICINAL O ENPRESENCIA DE ESTE.

PRECAUCIÓN: No coloque la unidad de manera que elacceso al cable de alimentación resulte complicado.

ADVERTENCIA: NO USE SU CONCENTRADOR DEOXÍGENO EN PRESENCIA DE GASES INFLAMABLES.
ESTO PUEDE GENERAR UN FUEGO RÁPIDOQUE PUEDE PROVOCAR DAÑOS A LOS BIENES,LESIONES CORPORALES O LA MUERTE.

PRECAUCIÓN: Al usar el concentrador de oxígeno
en un automóvil, embarcación o en otras fuentes deCC con la fuente de alimentación de CC, asegúrese
de que el vehículo esté funcionando antes de conectar
el concentrador de oxígeno. Si no se siguen estas
instrucciones, la fuente de alimentación podría nosuministrar energía al concentrador de oxígeno.

ADVERTENCIA: RETIRAR LA CÁNULA Y COLOCARLASOBRE LA ROPA, LA CAMA, EL SOFÁ U OTRO
MATERIAL ACOLCHADO CAUSARÁ UNA LLAMARADASI SE LA EXPONE A UN CIGARRILLO, UNA FUENTE
DE CALOR O UNA LLAMA.
ADVERTENCIA: NO DEJE UNA CÁNULA NASAL
SOBRE O DEBAJO DE LA ROPA, LA CAMA O LOSALMOHADONES DE UNA SILLA. SI LA UNIDAD
ESTÁ ENCENDIDA PERO NO EN USO, EL OXÍGENO
PODRÍA VOLVER INFLAMABLES ESOS MATERIALES.
ASEGÚRESE DE QUE LA ALIMENTACIÓN ESTÉAPAGADA CUANDO EL CONCENTRADOR DEOXÍGENO NO ESTÉ EN USO.
ADVERTENCIA: LOS ACCESORIOS DE SUMINISTRODE OXÍGENO APROBADOS ESTÁN DISEÑADOS
PARA REDUCIR LA PROPAGACIÓN DEL FUEGOPARA LA SEGURIDAD DEL PACIENTE. EL USO DE
ACCESORIOS NO APROBADOS PUEDE AUMENTAREL RIESGO DE LESIONES O DAÑOS RELACIONADOS
CON INCENDIOS.
ADVERTENCIA: EL CONCENTRADOR DE OXÍGENO
DEBE ESTAR APAGADO Y DESCONECTADO DE LA
ALIMENTACIÓN EXTERNA ANTES DE PROCEDER A
RETIRAR LA BATERÍA.
ADVERTENCIA: LAS PIEZAS PEQUEÑAS
PUEDEN CONSTITUIR UN RIESGOPOTENCIAL DE AHOGAMIENTO.
ADVERTENCIA: EVITE EL CONTACTO CON LA
SALIDA DE AIRE DE ESCAPE DURANTE MÁS
DE 10 SEGUNDOS, YA QUE PUEDE ESTAR CALIENTE.
ADVERTENCIA: EL VIENTO O LAS CORRIENTES
FUERTES PUEDEN INFLUIR NEGATIVAMENTEEN EL SUMINISTRO DE OXIGENOTERAPIA.
ADVERTENCIA: ESTE PRODUCTO PUEDE
EXPONERLO A SUSTANCIAS QUÍMICAS COMO
EL NÍQUEL, QUE, SEGÚN EL ESTADO DE
CALIFORNIA, CAUSA CÁNCER. PARA OBTENER MÁS
INFORMACIÓN, VISITE WWW.P65WARNINGS.CA.GOV.
PRECAUCIÓN: Siempre desconecte la fuentede alimentación de CA de la pared antes dedesconectar la fuente de alimentación de
CA del concentrador de oxígeno.

112 - ES PN MN232-C4 L | Manual del usuario

PRECAUCIÓN: Siempre coloque el tubo de suministrode oxígeno y los cables de alimentación de forma tal
que evite el riesgo de tropiezos o de estrangulación
accidental.

PRECAUCIÓN: Cuando el automóvil en el que se usa
la unidad del concentrador de oxígeno se apague,
desconecte y retire el dispositivo del automóvil.
No almacene el concentrador de oxígeno en un
automóvil muy caliente o muy frío o en otro ambiente
similar de alta o baja temperatura.
Aclaración: Los equipos de comunicaciones portátiles
y móviles de radiofrecuencia pueden afectar a losequipos eléctricos médicos.
PRECAUCIÓN: NO deje el concentradorde oxígeno ni la fuente de alimentación
enchufados en el vehículo si se encuentra
apagado. De lo contrario, podría agotar labatería del vehículo.
PRECAUCIÓN: Si el concentrador de oxígeno se
opera fuera de su intervalo de temperatura defuncionamiento normal y rango de presión ambiente(altitud), se puede ver afectado el rendimiento y
reducirse el tiempo de funcionamiento de la batería oaumentar el tiempo de carga de la batería. Consultela sección de especificaciones de este manual para
conocer los límites de temperatura de funcionamientoy almacenamiento.
PRECAUCIÓN: Para concentradores de oxígeno
equipados con baterías: Para garantizar la duraciónde la vida de la batería, manténgala almacenada en
un lugar fresco y seco. Guardar el concentradorde oxígeno durante un lapso prolongado a altas
temperaturas o con una batería totalmente cargada/descargada puede reducir la vida útil general de labatería. NO intente abrir la batería, ya que no disponede piezas que puedan repararse en su interior.
Mantenga las baterías lejos de los niños.
PRECAUCIÓN: SOLO USE las baterías suministradas
por el fabricante. Para la eliminación correcta de labatería, contacte a su proveedor del equipo o a suorganismo gubernamental local a fin de conocer losrequisitos de eliminación.

FreeStyle Comfort
Aclaración: Para mitigar los daños causados por elconcentrador o un fallo de alimentación, el proveedor delequipo debe evaluar la necesidad de oxígeno de reserva
durante la instalación y periódicamente cuando estos
atributos cambien:- El estado del paciente- El entorno en el que vive el paciente- La capacidad de reabastecer al paciente con
suministros de reserva de oxígeno suplementario
Aclaración: Asegúrese de que la cánula esté bien
insertada y quede firme. Esto asegura que elconcentrador de oxígeno pueda detectar apropiadamente
la inspiración para la entrega de oxígeno. Durante la
inhalación, debe escuchar o sentir el flujo de oxígeno
hacia el extremo de la cánula nasal.

Aclaración: Si el concentrador de oxígeno se ha
almacenado por un período prolongado fuera de su
estado de funcionamiento ambiente normal mostrado
en la tabla de especificaciones, permita que la unidad
vuelva a la temperatura de funcionamiento normal antes
de encenderla.
Aclaración: El proveedor del equipo debe evaluar lanecesidad de oxígeno de reserva en caso de fallo delconcentrador o del suministro eléctrico en el momentode la instalación y periódicamente cuando estos atributos
cambien:- El estado del paciente- El entorno en el que vive el paciente- La capacidad de reabastecer al paciente con
suministros de reserva de oxígeno suplementario

Aclaración: Las baterías de litio pueden perder su
capacidad para siempre si se exponen a temperaturas
extremadamente elevadas estando las baterías
totalmente cargadas o descargadas. Si se va a
almacenar por un periodo prolongado, se recomienda
cargar las baterías a un nivel entre el 25 % y 50 %de su capacidad y guardarlas a una temperatura de-20 °C-20 °C (-4 °F-68 °F).
Aclaración: No opere el concentrador de oxígeno sin
el filtro de entrada de aire colocado. Si dispone de
un segundo filtro, coloque este filtro de reemplazoantes de limpiar el filtro sucio. Limpie el filtro sucio con
una solución tibia de agua y jabón y luego seque porcompleto antes de usar.
Aclaración: El fabricante no recomienda la esterilizaciónde este equipo.
Aclaración: En ningún caso existe el peligro de reducir
el oxígeno presente en una habitación al usar elconcentrador de oxígeno.
ADVERTENCIA: EL CONCENTRADORDE OXÍGENO NO SE DEBE OPERARNI CONECTAR A UNA FUENTE DE
ALIMENTACIÓN EXTERNA (CA/CC)SIN HABER COLOCADO LA BATERÍAPREVIAMENTE. NO INTENTE ENCENDER
LA UNIDAD O CONECTARLA A UNAFUENTE DE ALIMENTACIÓN EXTERNA SI
SE HAN EXTRAÍDO LAS BATERÍAS.
RETIRE LA BATERÍA SOLO CUANDO LA UNIDADESTÉ APAGADA Y DESCONECTADA DE LA FUENTE
DE ALIMENTACIÓN. NO INTENTE RETIRAR LA
BATERÍA MIENTRAS LA UNIDAD ESTÉ ENCENDIDA OCONECTADA A LA FUENTE DE ALIMENTACIÓN.

PN MN232-C4 L | Manual del usuario ES - 113

FreeStyle Comfort
Especificaciones
FreeStyle Comfort
Dimensiones (con una sola batería)

25,4 x 18,5 x 7,9 cm (10,0 x 7,3 x 3,1 in)Dimensiones (con dos baterías)

27,9 x 18,5 x 7,9 cm (11,0 x 7,3 x 3,1 in)

Peso (con una sola batería)

2,3 kg (5 libras)

Peso (con batería doble)

2,7 kg (6 libras)

Nivel nominal del sonido***39,93 dB(A) ± 0,42 dB(A) @ 2

Ajustes de flujo

Ajuste de pulso 1, 2, 3, 4, 5

Ajuste de salida de O2

Configuración 1-210 ml/min
Configuración 2-420 ml/min
Configuración 3-630 ml/min
Configuración 4-840 ml/min
Configuración 5-1050 ml/min([@20bpm STPD] Todos los volúmenes +/- 15 %)Concentración de oxígeno*
Monitor de oxígeno

90 % (+5,5 % / -3 %)SíPresión máxima de salida<30 PSI

Sensibilidad de dosificación> -0,1 cm H2O (ajustable)

Energía de CA

100-240 V AC, 50-60 Hz, 120 W máx.

Energía de CC

11-18 V CC (10 A máx.)

Tipo de batería

iones de litio

Capacidad de la batería
Tiempo de recarga de la batería:
Duración de la batería
Temperatura de funcionamiento**

Celda única: 6700 mAh; doble celda: 13 400 mAh
Batería única: 3,5 horas
Batería doble: 6,0 horas
Duración de la batería única: Configuración 2: hasta 4 horas
Duración de la batería doble: Configuración 2: hasta 8 horas
41 °F a 104 °F (5 °C a 40 °C);

Temperatura de transporte y almacenamiento(Unidad)De -25 °C a +70 °C (de -13 °F a +158 °F) hasta 3048 m (10000 ft)

Temperatura de transporte y almacenamiento(batería)

De -25 °C a +70 °C (de -13 °F a +158 °F) hasta 3048 m (10000 ft)

Funcionamiento en altitud**De -381 a 3048 m (de -1250 a 10000 ft) (probado a 700-1060 hPa)

Humedad de funcionamiento

Humedad relativa de 15-95 % (sin condensación)

Humedad de almacenamiento
Garantía limitada

0-90 % sin condensación
Ver la declaración de garantía*Con base en el intervalo de presión atmosférica de 14,7 psi (101 kPa) a 21 °C (70 °F).
**La unidad funcionará en todos los entornos siempre que el entorno se encuentre dentro de los intervalos de temperatura
y humedad especificados. El funcionamiento fuera de estas especificaciones puede limitar la capacidad del concentradorde suministrar la especificación de concentración de oxígeno a velocidades de flujo de litro más altas. Es posible que sesiga visualizando la pantalla de calentamiento hasta que se alcance el nivel de concentración objetivo o pasen 15 minutos.
Continúe usando la unidad durante este tiempo.
***Nivel sonoro medido por el método de prueba n.º 14-1 10/2018 MDS-Hi.
La vida útil esperada es de un mínimo de cinco años.

114 - ES PN MN232-C4 L | Manual del usuario

FreeStyle Comfort
Instrucciones de funcionamiento
1. Localice y coloque el FreeStyle Comfort en una
posición adecuada para su uso.
a. El concentrador debe estar bien ventilado paraque los filtros de entrada y salida de aire no se
obstruyan.
b. Coloque el concentrador de modo que losindicadores visuales y sonoros y las alarmas
puedan verse y oírse fácilmente.
c. Asegúrese de que los filtros de entrada de aireestén bien colocados antes de utilizar la unidad.
Póngase en contacto con su proveedor delequipo si falta algún filtro de la unidad.
2. Instale la batería en el concentrador.
a. Alinee la batería con el alojamiento inferior delcentro del FreeStyle Comfort.

3. Inserte la fuente de alimentación de CA o CC en el
receptáculo del puerto de alimentación ubicado en laparte delantera inferior derecha de la unidad.

4. Conecte una cánula nasal al puerto de salida deoxígeno. Asegúrese de que la conexión sea segura
antes de poner el sistema en funcionamiento.
ACLARACIÓN: Usar una cánula diferente de lasrecomendadas puede influir negativamente en el
rendimiento.

b. Deslice la batería hacia dentro hasta que
escuche un clic para confirmar que se hayacolocado correctamente.

Aclaración: Al proveedor del equipo: se recomiendan lossiguientes accesorios para la administración de oxígeno
para usar con FreeStyle Comfort:• Cánula nasal con un tubo de 2,1 m (7 pies)(LPM máximo: 6): Número de pieza CAIRE 5408-SEQ• La información adicional sobre accesorios
recomendados está disponible en línea en
www.caireinc.com.
5. Para encender el FreeStyle Comfort, pulse elbotón de encendido y seleccione el ajuste de flujo
recetado usando los botones «aumentar» (+) o (-)«disminuir» flujo.

PN MN232-C4 L | Manual del usuario ES - 115

FreeStyle Comfort
Frecuencia
respiratoria
15
20
25
30
35
40
Se emitirá una breve alarma sonora para indicar que
el dispositivo se ha encendido.
Aclaración: Cuando se enciende el FreeStyle Comfort por
primera vez, el ajuste de flujo predeterminado es 2. Si secambia el ajuste, la unidad se encenderá con el ajustede flujo previo.
6. Coloque la cánula nasal en su cara siguiendo las
instrucciones del fabricante suministradas con la
cánula y respire con normalidad. Debe sentir unsoplo de oxígeno suministrado a través de la cánula
nasal cada vez que inhala.
7. Para cambiar el ajuste de flujo, pulse los botones«aumentar» (+) o «disminuir» (-) flujo hasta
conseguir la velocidad de flujo deseada. Es normal
escuchar un sonido diferente a medida que cambia
los ajustes de flujo.

1

Volumen del pulso no autoSat (ml)*Configuración
2
3
4
5

14,0
10,5
8,4
7,0
6,0
5,3

28,0
21,0
16,8
14,0
12,0
10,5

42,0
31,5
25,2
21,0
18,0
15,8

56,0
42,0
33,6
28,0
24,0
21,0

70,0
52,5
42,0
35,0
30,0
26,3

8. Apague el FreeStyle Comfort pulsando el botónde encendido.
Aclaración: La pantalla entrará en modo de suspensión
y se atenuará al cabo de 20 segundos. Pulse cualquierbotón para que la pantalla salga del modo de suspensión.
9. Para las características avanzadas, consulte
Instrucciones del menú Navegación

CARACTERÍSTICAS ESPECIALES
DEL PRODUCTO:
autoSAT®
La característica autoSAT patentada de CAIRE
permite que FreeStyle Comfort administre un volumen
constante de oxígeno (bolo) por respiración, hasta
un índice de respiración máximo. A medida que la
frecuencia respiratoria del usuario aumenta durante
actividades como caminar, la función autoSAT ajustala operación del dispositivo para mantener un volumen
de bolo establecido.
UltraSense®
Este sensible disparador de dosis en pulso detecta elcambio de presión durante la respiración y libera elpulso de oxígeno cuando inhala.

Frecuencia
respiratoria
15
20
25
30
35
40

Volumen del pulso autoSAT (ml)*Configuración
2
3
4
5

1
10,5
10,5
10,5
10,5
10,5
10,5

21,0
21,0
21,0
21,0
21,0
21,0

31,5
31,5
31,5
31,5
30,0
26,3

42,0
42,0
42,0
35,0
30,0
26,3

*La tolerancia es de +/- 15 %

116 - ES PN MN232-C4 L | Manual del usuario

52,5
52,5
42,0
35,0
30,0
26,3

autoDOSE®
En caso de producirse una «alarma por falta derespiración», esta función emitirá un pulso de oxígeno
a 20 respiraciones por minuto, con el ajuste de flujo
actual de la unidad, hasta que se resuelva la alarma.

FreeStyle Comfort
Fuentes de alimentación
El FreeStyle Comfort se puede alimentar de tres maneras:• Batería recargable• Energía de CA• Energía de CC
En las siguientes secciones tiene información de cómo
funciona cada uno de estos componentes.
ADVERTENCIA: USE SOLO LAS FUENTES
DE ALIMENTACIÓN INDICADAS EN ESTEMANUAL DE USUARIO. EL USO DE
FUENTES DE ALIMENTACIÓN Y BATERÍASQUE NO SEAN DE CAIRE PUEDE
OCASIONAR DAÑOS EN LA UNIDAD
Y UN FUNCIONAMIENTO INCORRECTO.

QUITAR LA BATERÍA

Presione el botón de extracción de la batería del
FreeStyle Comfort. Mientras MANTIENE PULSADO
este botón, deslice la batería hacia fuera retirándola delconcentrador.
ADVERTENCIA: EL CONCENTRADORDE OXÍGENO DEBE ESTAR APAGADOY DESCONECTADO DE UNA FUENTE DE
ALIMENTACIÓN EXTERNA ANTES DE
PROCEDER A RETIRAR LA BATERÍA.

BATERÍA

El sistema FreeStyle Comfort puede funcionar con labatería recargable que se suministra con el dispositivo.
No hay limitaciones de funcionamiento cuando elconcentrador funciona alimentado por la batería. Todaslas velocidades de flujo están disponibles cuando elconcentrador recibe la alimentación de la batería.

COMPROBAR LA CARGA DE LA BATERÍA
INSTALAR LA BATERÍA

Alinee la batería con el alojamiento inferior del centrodel FreeStyle Comfort. Deslice la batería hacia dentro
hasta que escuche un clic para confirmar que se hayacolocado correctamente.

Para comprobar la carga de la batería que está instalada
en el FreeStyle Comfort, consulte el icono de batería
y el porcentaje de batería restante de la pantalla.
80-100 %60-80 %

40-60 %

20-40 %20 %PN MN232-C4 L | Manual del usuario ES - 117

FreeStyle Comfort
Aclaración: Si la fuente de alimentación está conectada, se
mostrará un indicador de enchufe en lugar del porcentaje
de carga restante.
Para comprobar la carga de la batería cuando noestá instalada en el FreeStyle Comfort, colóquela enhorizontal y pulse el botón de prueba TEST. Las luces
del indicador del estado de batería (25-100 %) se
iluminan a la derecha del botón TEST para indicar el
nivel de carga de la batería.

CARGA INICIAL

La batería nueva suministrada con el sistema FreeStyle
Comfort no está cargada completamente cuando se
envía de fábrica. Antes de usarlo por primera vez, debe
cargar completamente la batería usando una fuente dealimentación de CA o CC.

INFORMACIÓN GENERAL DE CARGAEl FreeStyle Comfort recargará la batería siempre que
esté instalada en el concentrador y el concentrador esté
conectado a una fuente de alimentación de CA o CC
(salvo que esté en un avión).
La batería está totalmente cargada cuando se muestra
un 100 %.
Si lo desea, puede cargar la batería usando el cargador
de escritorio opcional.

118 - ES PN MN232-C4 L | Manual del usuario

DURACIÓN DE FUNCIONAMIENTODE LA BATERÍA

Cuando se desconecta una fuente de alimentación
CA o CC externa, el concentrador pasará a usar
automáticamente la alimentación de batería si hay una
batería cargada instalada. La siguiente tabla muestra las
duraciones típicas de una batería nueva. Las duraciones
de la batería son constantes, independientemente de su
frecuencia respiratoria.
Ajustede pulso

Duración de labatería única
(8 celdas)

Duración de labatería doble(16 celdas)

1

hasta 8 horas

hasta 16 horas

2hasta 4 horas

hasta 8 horas

3hasta 3 horas

hasta 6 horas

4hasta 2,25 horas

hasta 4,5 horas

5hasta 2 horas

hasta 4 horas

TIEMPO TÍPICO DE RECARGADE LA BATERÍA

El tiempo típico de recarga de una bateríacompletamente descargada es de 3,5 horas para una
batería única y de 6,0 horas para una batería doble.
Aclaración: Es posible que la batería no empiece a cargar
inmediatamente si está completamente descargada.

VIDA ÚTIL DE LA BATERÍA

Las baterías del FreeStyle Comfort están diseñadas
para durar 300 ciclos de carga/descarga.

FreeStyle Comfort
Fuente de alimentación de CA y CC
El FreeStyle Comfort viene con una fuente dealimentación de CA y CC estándar.
La fuente de alimentación de CA incluye lo siguiente:	a. El cargador de CA que funciona entre 100
y 240 voltios, 50-60Hz.
	b. El cable de alimentación de CA para conectar
al cargador y salida de CA correspondiente.

La fuente de alimentación de CC consta de un cableque se enchufa directamente en la unidad y en la salida
de CA.

2. Conecte el cable de alimentación de CA aladaptador de CA.

ENERGÍA DE CA

La energía de CA universal se usa en el hogar o donde
esté disponible una salida de alimentación de CA.
El FreeStyle Comfort es totalmente funcional con laalimentación de CA. El FreeStyle Comfort también
carga la batería mientras recibe alimentación de CA(excepto en un avión). La batería se cargará tanto si el
FreeStyle Comfort está encendido como apagado.
ADVERTENCIA: NO CONECTE
ALIMENTACIÓN EXTERNA SIN HABER
COLOCADO PREVIAMENTE LA BATERÍA
EN EL CONCENTRADOR DE OXÍGENO.
Para conectar el FreeStyle Comfort a la alimentaciónde CA:
1. Conecte el adaptador de CA a la entrada delconector de alimentación ubicado en la partedelantera inferior derecha del FreeStyle Comfort.
Asegúrese de empujarla hacia dentro para que laconexión sea segura.

3. Conecte el extremo opuesto del cable dealimentación de CA a la toma de pared o a otra
salida eléctrica adecuada.

PN MN232-C4 L | Manual del usuario ES - 119

FreeStyle Comfort
4. Para encender el FreeStyle Comfort, pulse elbotón de encendido y seleccione el ajuste de flujo
recetado usando los botones «aumentar» (+) o (-)«disminuir» flujo.

Aclaración: Si se necesita cargar la batería, el icono
de batería mostrará una única barra de color naranja.

ENERGÍA DE CC

La energía de CC se usa en salidas para accesorios devehículos, embarcaciones y otros vehículos a motor.
El FreeStyle Comfort es totalmente funcional con laalimentación de CA. La batería del FreeStyle Comfort
también se carga mediante la alimentación de CC. La
batería se carga tanto con la unidad apagada comoencendida y en uso.
ADVERTENCIA: NO CONECTE
ALIMENTACIÓN EXTERNA SIN HABER
COLOCADO PREVIAMENTE LA BATERÍA
EN EL CONCENTRADOR DE OXÍGENO.
1. Arranque el motor del vehículo.
2. Conecte la fuente de alimentación de CC al
receptáculo del puerto de alimentación ubicado enla parte delantera inferior derecha del FreeStyle
Comfort. Asegúrese de empujarla hacia dentro paraque la conexión sea segura.

4. Para encender el FreeStyle Comfort, pulse elbotón de encendido y seleccione el ajuste de flujo
recetado usando los botones «aumentar» (+) o (-)«disminuir» flujo.
PRECAUCIÓN: Para evitar un posible
incendio o sobrecalentamiento, asegúrese
de que la toma de corriente de 12 V CC
esté libre de residuos y de que el enchufe
adaptador de la fuente de alimentación de
CC encaje correctamente y permanezca
bien colocado.
PRECAUCIÓN: Las superficies metálicas de la fuentede alimentación de CC pueden CALENTARSE MUCHO
cuando se utilizan debido a la alta corriente. Deje
tiempo suficiente para que las superficies se enfríen
antes de tocarlas.
PRECAUCIÓN: No utilice la fuente de alimentación deCC con extensiones, divisores o adaptadores. El uso
de accesorios no homologados puede provocar unsobrecalentamiento o daños.
PRECAUCIÓN: Conectar el dispositivo a una tomade corriente CC de 12 V puede agotar la batería delvehículo si el motor no está en marcha, aunque eldispositivo esté apagado.
PRECAUCIÓN: No arranque un vehículo mientras lafuente de alimentación de CC esté conectada.

3. Conecte el extremo opuesto del cable dealimentación de CC a la salida del vehículo o acualquier otra salida para accesorios de CC.

120 - ES PN MN232-C4 L | Manual del usuario

FreeStyle Comfort
Limpieza, cuidados ymantenimiento periódico
MANTENIMIENTO PERIÓDICO

ADVERTENCIA: NO OPERE EL CONCENTRADOR DE OXÍGENO SIN EL FILTRO DEENTRADA DE AIRE COLOCADO.

CUIDADO DE LA BATERÍA

Mantenimiento Preventivo
El dispositivo no requiere mantenimiento. Los filtros
dentro del dispositivo están pensados para durar
toda la vida útil del dispositivo. Si el dispositivo no
funciona correctamente, debe devolverse la unidad alproveedor para su reparación. El mantenimiento delos componentes internos del gabinete del FreeStyle
Comfort ha de ser llevado a cabo por el proveedor delconcentrador de oxígeno.

No permita que la batería quede expuesta a líquidos.

El operador de la unidad es responsable del siguiente
procedimiento de mantenimiento.
• Semanalmente: Limpiar los filtros de entrada de aire

Si parece que la batería está dañada, póngase en
contacto con su proveedor del concentrador de oxígeno
antes de usarla.

ADVERTENCIA: NO INTENTE ABRIR O
RETIRAR LAS CUBIERTAS DE PLÁSTICODEL CONCENTRADOR DE OXÍGENO.
LA UNIDAD SOLO DEBE SER ABIERTAPOR UN TÉCNICO DE MANTENIMIENTOCUALIFICADO.

FILTROS DE ENTRADA DE AIRE

El aire entra al FreeStyle Comfort a través de filtros
de entrada de aire ubicados en la parte inferior frontaldel concentrador de oxígeno. Los filtros evitan que elpolvo y otras partículas grandes existentes en el aire
entren a la unidad.
Debe limpiar los filtros de entrada de aire una vez
por semana como mínimo. Puede que su proveedordel concentrador de oxígeno le aconseje limpiarlo
con más frecuencia dependiendo de las condiciones
de funcionamiento. Siga estos pasos para limpiar
correctamente los filtros de entrada de aire.
1. Extraiga los dos filtros de la parte frontal del
FreeStyle Comfort. Para ello, presione la pestaña
correspondiente y tire de ella.

No intente desmontar ni abrir la batería.
No exponga la batería a llamas abiertas.
No deje caer la batería.
Mantenga la batería fuera del alcance de los niños.

No coloque la batería dentro ni cerca de un microondas
ni ningún otro aparato de cocina.

ALMACENAMIENTO DE LA BATERÍA

No es necesario que la batería del concentradorde oxígeno esté completamente descargada antes
de volver a cargarla. Se recomienda cargar la bateríadel concentrador de oxígeno después de cada uso.
No deje la batería en un lugar expuesto a calor
excesivo, como luz directa solar o en un coche.
De lo contrario, el rendimiento y la vida útil de labatería podrían deteriorarse. También puede hacer quela batería genere calor, humo, prenda fuego o explote.
La batería se debe almacenar en un entorno estable
caracterizado por una baja humedad (humedad relativa
inferior al 70 %) donde no haya gases corrosivos. Para
evitar la oxidación, evite condiciones que puedan crearcondensación tales como fluctuaciones rápidas detemperatura ambiente.
Para guardarla durante un mes o más, se recomienda
que la batería esté descargada o cargada entre el 50 %y 75 % (de dos a tres LED). Una vez cargada, extraiga
la batería del concentrador de oxígeno.

2. Lave ambos filtros con una solución templadade agua y jabón.
3. Enjuague bien los filtros y elimine el excesode agua con un paño suave y absorbente. Asegúrese
de que los filtros estén completamente secos antes
de volver a colocarlos.
4. Vuelva a colocar los filtros limpios y secos.
PN MN232-C4 L | Manual del usuario ES - 121

FreeStyle Comfort
LIMPIEZAAPAGUE la unidad y desconéctela de la toma de CAo de CC antes de limpiarla. NO pulverice la carcasa
exterior directamente. Utilice un paño húmedo
(no mojado) o una esponja. Pulverice el paño o laesponja con una solución limpiadora doméstica suave
para limpiar el gabinete y la fuente de alimentación.
Siga las instrucciones del fabricante del limpiador.
ADVERTENCIA: APAGUE LA UNIDADY DESCONECTE EL CABLE DE
ALIMENTACIÓN DE LA TOMA ELÉCTRICAANTES DE LIMPIAR LA UNIDAD PARAEVITAR UNA DESCARGA ELÉCTRICA
ACCIDENTAL O EL RIESGO DE
QUEMADURAS. ÚNICAMENTE ELPROVEEDOR DE SU EQUIPO O UNTÉCNICO DE SERVICIO CUALIFICADODEBEN QUITAR LAS TAPAS O REALIZARLA INSPECCIÓN DE LA UNIDAD.
ADVERTENCIA: HAY QUE TENER CUIDADO DE
EVITAR QUE EL CONCENTRADOR DE OXÍGENOY LA BATERÍA SE MOJEN Y NO DEBE PERMITIRSEQUE NINGÚN LÍQUIDO ENTRE EN LA UNIDAD. ESTO
PUEDE CAUSAR UN FALLO O APAGADO Y PRODUCIRUN MAYOR RIESGO DE DESCARGA ELÉCTRICAO QUEMADURAS.
ADVERTENCIA: NO USE LÍQUIDO EN CONTACTODIRECTO CON LA UNIDAD. A CONTINUACIÓN,SE INCLUYE UNA LISTA DE ALGUNOS AGENTESQUÍMICOS NO DESEABLES, ENTRE OTROS:
ALCOHOL Y PRODUCTOS A BASE DE ALCOHOL,PRODUCTOS A BASE DE CLORO CONCENTRADO(CLORURO DE ETILENO) Y PRODUCTOS A BASE DE
ACEITE. ESTOS NO DEBEN USARSE PARA LIMPIAR
LA CARCASA DE PLÁSTICO DEL CONCENTRADORDE OXÍGENO, YA QUE PUEDEN DAÑAR LA UNIDAD.
ADVERTENCIA: LIMPIE EL GABINETE, EL PANEL DE
CONTROL Y EL CABLE DE ALIMENTACIÓN SOLOCON UN LIMPIADOR DOMÉSTICO APLICADO CON
UN PAÑO HÚMEDO (NO MOJADO) O UNA ESPONJAY LUEGO SEQUE TODAS LAS SUPERFICIES. NO
PERMITA QUE NINGÚN LÍQUIDO SE INTRODUZCA EN
EL CONCENTRADOR. PRESTE ESPECIAL ATENCIÓN
A LA SALIDA DE OXÍGENO PARA LA CONEXIÓN
DE LA CÁNULA PARA ASEGURARSE DE QUE SE
MANTENGA LIBRE DE POLVO, AGUA Y PARTÍCULAS.

122 - ES PN MN232-C4 L | Manual del usuario

ADVERTENCIA: UTILICE UN PAÑO HÚMEDO (NOMOJADO) O UNA ESPONJA PARA LIMPIAR LA
BATERÍA. PRIMERO, PULVERIZE EL PAÑO O LAESPONJA CON UN DETERGENTE DOMÉSTICO SUAVEY LIMPIE EL ALOJAMIENTO DE LA BATERÍA.
ADVERTENCIA: NO LIMPIE CON UN SISTEMA DE
LAVADO AUTOMATIZADO.

REEMPLAZO DE LA CÁNULA

Siempre siga las instrucciones del fabricante de lacánula para un uso correcto. Reemplace la cánula
nasal o el tubo de oxígeno cada periodo de entre 3 y
6 meses (si se usa a diario, cada 3 meses) atendiendoa la recomendación del fabricante de la cánula o desu proveedor del concentrador de oxígeno. El médico
o proveedor del concentrador de oxígeno le facilitarán
las instrucciones de limpieza y reemplazo de laspiezas.
El proveedor del concentrador de oxígeno dispone
de suministros adicionales de piezas de repuesto.

FreeStyle Comfort
Estados de alarma
N.° deprioridad

Descripciónde la alarma

1

Fallo delsistema
por alta
temperaturade batería

Pantalla

Señal
sonoraMás detalles / Acción sugerida

1 pitido
cada 20 s

La unidad emitirá una alarma cuando latemperatura de la alarma esté alcanzando
el umbral de apagado de seguridad.
Si se alcanza el umbral de apagado, la unidad
se apagará. Si se conecta la alimentación
externa, se mostrará una pantalla de alarma
y sonará una alarma durante 2 minutos.
Probar con batería a temperatura ambiente.
Si el problema persiste, póngase en contacto
con el proveedor.

2

Fallo delsistema
por alta
temperatura

1 pitido
cada 20 s

La unidad emitirá una alarma cuando
la temperatura interna esté alcanzando
el umbral de apagado de seguridad.
Si se alcanza el umbral de apagado, semostrará una pantalla de alarma y sonará una
alarma durante 2 minutos.
Asegúrese de que no haya obstrucciones en
ninguna de las aberturas de la unidad. Si el
problema persiste, póngase en contacto conel proveedor.

3

Fallo delsistema por
alta presión(alta presióndel sistema
o fallo delcompresor)

1 pitido
cada 20 s

4

Fallo delsistema por
velocidad
de flujo(fallo de laválvula OCD,
obstrucción dela ruta del gas)

1 pitido
cada 30 s

Si el problema persiste, póngase en contacto
con el proveedor.

5

Apagado porbatería baja

1 pitido
cada 30 s

Emite una alarma hasta que la batería
se agota y la unidad se apaga.

Si se alcanza el umbral de apagado,
se mostrará una pantalla de alarma
y sonará una alarma durante 2 minutos.
Si el problema persiste, póngase en contacto
con el proveedor.

Cargue la batería.

PN MN232-C4 L | Manual del usuario ES - 123

FreeStyle Comfort
N.° deprioridad

Descripciónde la alarma

6

No se detecta
respiración

Pantalla

Señal
sonoraMás información

1 pitido
cada
10 minNo se detecta respiración durante 15 segundos.
Se activa cuando la pantalla de primera
respiración o calentamiento se apaga,lo que ocurra primero.
Compruebe que no haya presiones o fugas
en la cánula. Aumente la configuración desensibilidad. Si el problema persiste, póngase
en contacto con el proveedor.

7

Calentamiento

ningunoLa unidad iniciará la dosificación automática
hasta que se detecte la respiración o finaliceel calentamiento, lo que ocurra primero.

8

Bajaconcentración

1 pitido
cada
10 min,duranteunmáximode 30 minSi el problema persiste, póngase en contacto
con el proveedor.

9

Advertencia debatería baja

1 pitido,
sinrepetición

Cargue la batería.

10

Frecuencia
respiratoria
excedida

1 pitido,
sinrepetición

La frecuencia respiratoria excede los 40 bpm.

11

Fallo delsistema por
avería delventilador

ningunoSi el problema persiste, póngase en contacto
con el proveedor.

12

Fallo delsistema por
baja presión

ningunoSi el problema persiste, póngase en contacto
con el proveedor.

13

Fallo del sistema por baja
temperatura

ningunoDeje que la unidad alcance la temperatura
ambiente. Si el problema persiste, póngase
en contacto con el proveedor.

124 - ES PN MN232-C4 L | Manual del usuario

En caso de que la dosis no se suministre
correctamente, disminuya el ajuste desensibilidad. Si el problema persiste,póngase en contacto con el proveedor.

FreeStyle Comfort
N.° deprioridad

Descripciónde la alarma

Info

Arranque

Pantalla

Señal
sonoraMás información

1 pitido
informativo

La pantalla se enciende durante el arranquey debe escuchar un pitido para confirmar que
el timbre funciona
La pantalla de arranque muestra la versiónde firmware en la parte inferior derecha de laesquina.

Info

Pantalla defuncionamiento normal enmodo autoSAT(no hay una
fuente dealimentación
CC conectada)ningunoInfo

Pantalla defuncionamiento normal con
modo autoSAT
deshabilitado(no hay una
fuente dealimentación
CC conectada)ningunoInfo

Ajuste cambiado

1 pitido
informativo

Info

Alimentación
externa
conectada
mientras está
apagadoningunoInfo

Conecte laalimentación
externamientras estáen funcionamiento

ningunoInfo

Desconecte laalimentación
externamientras está
apagado1 pitido
informativo

Se muestra la pantalla CAIRE durante
2 segundos, seguido del estado de la batería
y del cable durante 5 s antes de que lapantalla se apague

Se muestra la pantalla CAIRE durante
2 segundos y luego la pantalla se apaga.

PN MN232-C4 L | Manual del usuario ES - 125

FreeStyle Comfort
N.° deprioridad

Descripciónde la alarma

Pantalla

Info

Desconecte laalimentación
externamientras
está en funcionamiento(con batería)

1 pitido
informativo

Info

En modo desuspensión:
cuando se
pulsa el botónde arriba oabajo la pantalla muestra
la batería yel estado decarganingunoInfo

Modo deavión enalimentaciónde batería

1 pitido
largo

Info

Modo de avión
mientras estáenchufado

1 pitidolargo

126 - ES PN MN232-C4 L | Manual del usuario

Señal
sonoraMás información

La pantalla se iluminará durante 5 s
y después se apagará

FreeStyle Comfort
Instrucciones del menú Navegación
Es posible acceder al menú en modo de suspensióno de funcionamiento. En el modo de funcionamiento,
en caso de que se produjese una alarma, se saldrádel menú automáticamente. Si durante 20 segundos
no se presiona ningún botón, se saldrá del menú
automáticamente. En el modo de suspensión, es
posible que tenga que presionar el botón Menú dos
veces para que la unidad salga del modo de suspensión
y poder entrar en el menú.
Hay cuatro pantallas Menú:

Modo avión

1. Pantalla de estado: Muestra el número de versiónde firmware, el tiempo de funcionamiento,
los ciclos de descarga
2. Modo avión (encendido/apagado)
3. Sensibilidad: ajustable (1-5)
4. autoSAT (encendido/apagado)

Sensibilidad

Ejemplo de pantalla de estado

FUNCIONAMIENTO DEL MENÚ AJUSTES

autoSATEl menú Ajustes se introduce al presionar el botón
Menú/Seleccionar
en los modos defuncionamiento o de suspensión.
Para navegar por los menús, pulse.

Página de estado

PN MN232-C4 L | Manual del usuario ES - 127

FreeStyle ComfortPÁGINA DE ESTADO
La página de estado del Freestyle Comfort muestra larevisión del firmware, las horas de funcionamiento y elnúmero de ciclos de batería de la batería conectada.

Para cancelar el modo avión, presione
el icono CANCEL esté visible.

mientras

Modo avión encendido confirmado
Página de estado

Las horas de funcionamiento se muestran en horas
enteras.
Las descargas de la batería son el número de ciclos quela batería adjunta se ha drenado y recargado.

MODO AVIÓNEl FreeStyle Comfort se envía con el modo avión
apagado.

Para apagar el modo avión, repita los pasos anteriores
mientras se coloca en «OFF».
Al presionar
cuando la pantalla muestra elsímbolo «Next», el menú pasará a la siguiente opciónde pantalla.
El modo avión está diseñado para usarse en modo defuncionamiento. Se deshabilitará automáticamente si launidad se deja en suspensión con la batería.
Aclaración: Consulte la sección Activar modo avión paraobtener más información sobre el modo avión.

MODO DE SENSIBILIDAD

Pantalla principal del modo avión

El FreeStyle Comfort se envía desde fábrica con dos
ajustes del modo de sensibilidad predeterminados y no
debería necesitar de ajustes en condiciones normales.
El ajuste de sensibilidad se puede ajustar entre losvalores 1 y 5, donde 1 es el menos sensible y 5 el más
sensible.

Para encender el modo, presione
mientrasel icono CHANGE esté visible. El valor pasaráa «ON», en azul.

Pantalla del modo de sensibilidad con el ajuste 2

Modo avión encendido seleccionado

Para confirmar el modo avión, presione
mientrasel icono SAVE esté visible. El valor pasará a «ON»,en negro.

128 - ES PN MN232-C4 L | Manual del usuario

FreeStyle Comfort
Para ajustar el modo de sensibilidad, presione
mientras el icono CHANGE esté visible. El valor
pasará al siguiente número azul.

Ajuste 3 del modo de sensibilidad seleccionado

Para confirmar el ajuste 3 del modo de sensibilidad,
presione
mientras el icono SAVE esté visible.
El valor pasará a «3», en negro.

MODO AUTOSATEl FreeStyle Comfort se envía con el modo autoSAT
encendido.

Pantalla principal del modo autoSAT

Para apagar el modo, presione
mientras el icono
CHANGE esté visible. El valor pasará a «OFF», en azul.

Para cancelar el modo de sensibilidad, presione
mientras el icono CANCEL esté visible.

Modo autoSAT apagado seleccionado

Ajuste 3 del modo de sensibilidad confirmado

Para confirmar el modo autoSAT, presione
mientras el icono SAVE esté visible. El valor pasaráa «OFF», en negro.
Para cancelar el modo autoSAT, presione
el icono CANCEL esté visible.

mientras

Ajuste 5 del modo de sensibilidad (máximo) confirmado

Al presionar
cuando la pantalla muestra elsímbolo «Next», el menú pasará a la siguiente opciónde pantalla.

Modo autoSAT apagado confirmado

Para encender el modo autoSAT, repita los pasos
anteriores mientras se coloca en «ON».
Al presionar
cuando la pantalla muestra elsímbolo «Next», el menú pasará a la siguiente opciónde pantalla.

PN MN232-C4 L | Manual del usuario ES - 129

FreeStyle Comfort
Solución de problemas
Problema

Causa probable

Solución

El sistema FreeStyle Comfort
hace sonar una alarma y emite
un pitido intermitente.

Consulte la tabla de Estadosde alarma.

Consulte la tabla de Estados de alarma.

El concentrador de oxígeno
no se enciende.

La batería está descargada.

Conecte la unidad a una fuente de alimentaciónde CA o CC.

No está conectado a una fuentede alimentación externa.
Falla general.

Asegúrese de que las conexiones externas
son seguras.
Haga un ciclo completo: retire la batería ydesconecte cualquier fuente de alimentación
externa CA o CC. A continuación, vuelva a
colocar la batería: el ciclo de alimentación se hacompletado. Si lo desea, vuelva a conectar lafuente de alimentación externa.
Comuníquese con el proveedor de su equipo
y cambie a otra fuente de oxígeno según sea
necesario.

Todos los demás problemas.

Haga un ciclo completo: retire la batería ydesconecte cualquier fuente de alimentación
externa CA o CC. A continuación, vuelva a
colocar la batería: el ciclo de alimentación se hacompletado. Si lo desea, vuelva a conectar lafuente de alimentación externa.
Cambie a otra fuente de oxígeno que haya disponibley póngase en con su proveedor del equipo.

Aclaración: Algunas alarmas harán que la unidad se apague.
Aclaración: Cuando se pierde la alimentación, el sistema de alarma fuerza un reinicio (elimina cualquier alarma anterior que
pueda haber) y solo se mostrará la alarma si la incidencia persiste después de que se haya restablecido la alimentación.

130 - ES PN MN232-C4 L | Manual del usuario

FreeStyle Comfort
Información de viaje
El FreeStyle Comfort cumple las directrices de la FAA
para usarse a bordo de aviones comerciales. Antes decualquier viaje (avión, tren, crucero, etc.), debe:• Asegurarse de que el concertador funcione
debidamente.
• Ponerse en contacto con su aerolínea al menos
48 horas antes de la salida del vuelo para informarlesde que viajará con un sistema FreeStyle Comfort.
Se recomienda viajar siempre con baterías adicionales.
Si va a viajar en avión, lleve la batería suficiente: tenga
en cuenta el tiempo en tierra (facturación, controles deseguridad, etc.) y escalas. La mayoría de las aerolíneas
requieren que cuente con un número adecuado debaterías cargadas completamente para alimentar eldispositivo por al menos 150 % del tiempo máximo deduración prevista del viaje (tiempos de vuelo, todo eltiempo en tierra, conexiones y demoras imprevistas).

1. Encienda el concentrador de oxígeno móvil SOLO
cuando no esté conectado a ninguna fuente dealimentación externa.

2. Habilite el modo avión mediante uno de estos
métodos:a. Siga las instrucciones para habilitar el modo
avión en la sección Instrucciones del menú
Navegación.

Aclaración: Si va a volar, es recomendable que llevebaterías adicionales en una bolsa de transporte segura.

REGLAMENTO DE LA FAA

Las baterías de repuesto deben protegerse
individualmente para evitar cortocircuitos, colocándolasen su embalaje de venta original o mediante terminalesde aislamiento; por ejemplo, encintando los terminales
expuestos o colocando cada batería en una bolsa deplástico o estuche protector por separado, y transportadas
en equipaje de mano solamente.

b. Presione y manténgalos presionados los botones
de arriba y abajo durante 5 segundos hasta que oiga
un pitido largo y vea que aparece al símbolo delavión.

ACTIVAR EL MODO AVIÓNAlimentación de batería
El concentrador de oxígeno móvil se puede usar
en una aeronave siguiendo los procedimientos defuncionamiento normales usando ÚNICAMENTE laalimentación de la batería, cuando no haya conectada
ninguna fuente de alimentación externa.
Alimentación externa
En un avión, si dispone de una salida eléctrica que
funcione y esté cerca de usted para usarla como fuentede alimentación del concentrador de oxígeno móvil,puede servirle a modo de reserva de las baterías. En
este caso, debe seguir el procedimiento recomendado
de transición de la batería a la alimentación eléctrica
de la aeronave:

3. Cuando se muestre el símbolo del avión, conecte elconcentrador de oxígeno móvil.

Aclaración: Debido a las limitaciones de las aeronaves,las baterías no se cargan durante el modo avión.

PN MN232-C4 L | Manual del usuario ES - 131

FreeStyle Comfort
Accesorios

INSERTAR LA UNIDAD EN EL BOLSO DE
TRANSPORTE

Para obtener un rendimiento y seguridad adecuados,
utilice solo los accesorios indicados en la tabla que
CAIRE suministra a través del proveedor de su
concentrador de oxígeno. El uso de accesorios que
no se mencionan a continuación podría afectar demanera negativa al rendimiento o a la seguridad delconcentrador.

1. Coloque la unidad en el fondo del bolso, alineando
adecuadamente los filtros y las aberturas de labatería.

Accesorios estándares del FreeStyle Comfort
Bolso de transporte

MI396-1

MochilaMI459-1

Correa para hombro

MI397-1

Fuente de alimentación de CA (todos
los cables incluidos)

PW036-1S

Fuente de alimentación de CC

CD041-1

Filtro de partículas externas grandes(derecha)FI226-1

Filtro de partículas gruesas externas(izquierda)

FI227-1

2. Asegure la cremallera alrededor de la parte inferiordel bolso.

Accesorios opcionales disponibles
Paquete de batería de 8 celdas
adicional

BT034-1S

Batería de 16 celdas

BT035-1S

Cargador de batería de escritorio

BT036-1

Bolsa para llevar todos los accesorios

MI372-2

Aclaración: puede haber opciones adicionales
disponibles para cables de alimentación específicos delpaís que no se mencionan arriba. Póngase en contacto
con CAIRE o con el proveedor de su concentrador deoxígeno si necesita encargar opciones alternativas.

Instrucciones del bolso detransporte
Para insertar y fijar su FreeStyle Comfort en el bolso
de transporte, siga las siguientes instrucciones.

132 - ES PN MN232-C4 L | Manual del usuario

COLOCACIÓN DE LA CORREA PARAHOMBRO
1. La correa para hombro se puede acoplar a la unidad
directamente o al bolso de transporte.
2. Acople la correa al aro de la unidad o del bolso.

FreeStyle Comfort
Eliminación y reutilización
Devuelva siempre a su proveedor del equipo el
FreeStyle Comfort junto con todos sus componentes
para su correcta eliminación. También puede ponerse
en contacto con las oficinas pertinentes de su
ciudad para obtener instrucciones sobre la correcta
eliminación de la batería. La unidad puede reutilizarse
entre personas, pero debe devolverse al proveedor delequipo para su reprocesamiento.

RAEE y RoHS
Este símbolo pretende recordar a los dueños
de los equipos que deben llevarlos a un centro dereciclado al final de su vida útil, de conformidad con
la Directiva sobre residuos de aparatos eléctricos
y electrónicos (RAEE).

Materiales en contacto directo con
el operador• Carcasa del concentrador: PC/ABS• Panel de control del concentrador: Poliéster EBG7o equivalente• Salida de oxígeno, boquilla: Aluminio• Etiqueta de la unidad: Lexan• Conectores de cable: Policarbonato/Cloruro de vinilo• Cable de alimentación: PVC, metal• Fuente de alimentación: Lexan 940 (policarbonato)• Batería: PC/ABS• Batería, etiquetas de suministro de alimentación
Película de poliéster
• Funda de transporte del concentrador:
Forro de cordura, neopreno, nailon• Correa: Polipropileno, acetil, vinilo, cordura,
neopreno

Nuestros productos cumplirán con la Restricción sobre
Sustancias Peligrosas (RoHS). No contendrán más que
trazas de plomo o de otros materiales peligrosos.
PRECAUCIÓN: Para un desecho adecuado,
comuníquese con su proveedor de equipos
o su agencia de gobierno local para conocer las instrucciones para el desecho.
PRECAUCIÓN: Elimine siempre el tamiz molecular deacuerdo con todas las normativas federales, estatales
y locales de control medioambiental.

PN MN232-C4 L | Manual del usuario ES - 133

FreeStyle Comfort
Prueba de EMC
Orientación y declaración del fabricante: emisiones electromagnéticas
El FreeStyle Comfort está destinado para su uso en el ambiente electromagnético que se especifica a continuación.
El cliente o el usuario del FreeStyle Comfort deben asegurar que se use en tal ambiente.
Prueba de emisiones

Cumplimiento

Ambiente electromagnético: orientación

Emisiones deradiofrecuencia

Grupo 1El sistema FreeStyle Comfort usa energía de radiofrecuencia solo parasu función interna. En consecuencia, sus emisiones de radiofrecuencia
son muy bajas y no es probable que causen interferencias en equipos
electrónicos cercanos.

CISPR 11
Emisiones deradiofrecuencia
CISPR 11
Emisiones armónicas
IEC 61000-3-2
Fluctuaciones de voltaje/emisiones de flicker
IEC 61000-3-3

Clase B
Clase A
Cumple134 - ES PN MN232-C4 L | Manual del usuario

El sistema FreeStyle Comfort es adecuado para usar en todos losestablecimientos, incluidos los establecimientos domésticos y aquellos
conectados directamente a la red pública de energía eléctrica de bajo
voltaje que alimenta edificios usados para fines domésticos.

FreeStyle Comfort
Orientación y declaración del fabricante: inmunidad electromagnéticaEl FreeStyle Comfort está destinado para su uso en el ambiente electromagnético que se especifica a continuación. El
cliente o el usuario del FreeStyle Comfort debe asegurarse de que se use en tal ambiente.
Prueba de inmuNivel de prueba IEC
Nivel de cumplimiento Ambiente electromagnético: orientación
nidad
60601
Ambiente
Los suelos deben ser de madera, hormigón
Contacto de ± 8 kV
Contacto de ± 8 kVelectromagnético:o cerámica. Si los suelos están cubiertos
Descarga de aire
Descarga de aire
orientación
con material sintético, la humedad relativa
de ± 15 kV
de ± 15 kV
debe ser de al menos un 30 %.
IEC 61000-4-2
Ráfagas eléctricas
de transitorios muy
rápidos
IEC 61000-4-4

Sobrecarga
IEC 61000-4-5

Caídas de voltaje,interrupciones
por cortocircuitos
y variaciones devoltaje en las
líneas de entrada
de las fuentes
de alimentación
IEC 61000-4-11± 2 kV para las líneas
de alimentación± 1 kV para líneas
de entrada/salida± 1 kV línea(s) a línea(s)±2 kV modo comúnen líneas de CA± 2 kV para las líneas
de alimentación
N/A± 1 kV línea(s) a línea(s)±1 kV diferencial enlíneas de CA

La calidad de la red eléctrica debe ser la deentornos de centros sanitarios y atención
sanitaria doméstica.

La calidad de la red eléctrica debe ser la deentornos de centros sanitarios y atención
sanitaria doméstica.

±2 kV modo comúnen líneas de E/S deexteriores
0 % UT para 0,5 ciclo
0 % UT para 0,5 ciclo(0°, 45°, 90°, 135°, 180°, (0°, 45°, 90°, 135°, 180°,
225°, 270°, 315°)
225°, 270°, 315°)

La calidad de la red eléctrica debe serla de entornos de centros sanitarios yatención sanitaria doméstica. Si el usuario
0 % UT para 1 ciclo (0°)
0 % UT para 1 ciclo (0°)
del sistema FreeStyle Comfort requiere un
funcionamiento continuo durante cortes
70 % UT (30 % caída de
70 % UT (30 % caída de de la red eléctrica, se recomienda que eltensión en UT) para 25/30 tensión en UT) para 25/30 FreeStyle Comfort se alimente desde una
ciclos (0°)
ciclos (0°)
fuente de alimentación sin interrupciones
o desde una batería.
0 % UT para 250/300
ciclos (0°)

0 % UT para 250/300
ciclos (0°)

Los campos magnéticos a frecuencia
industrial deben tener un nivel
característico de una ubicación típica en un
3 A/m
3 A/m
entorno de centro sanitario o de asistencia
sanitaria doméstica.
NOTA: UT es el voltaje de CA de la red eléctrica antes de la aplicación del nivel de prueba.

Campo magnético
a frecuencia
industrial (50/60 Hz)
IEC 61000-4-8

NOTA: En algunos casos, el FreeStyle Comfort puede restablecerse después de una descarga electrostática notable, y es
posible que necesite hacer un ciclo completo de alimentación para reanudar el funcionamiento normal. Consulte la secciónde resolución de problemas para obtener más información.

PN MN232-C4 L | Manual del usuario ES - 135

FreeStyle Comfort
Orientación y declaración del fabricante: inmunidad electromagnéticaEl FreeStyle Comfort está destinado para su uso en el ambiente electromagnético que se especifica a continuación.
El cliente o el usuario del FreeStyle Comfort debe asegurarse de que se use en tal ambiente.
Nivel de cumplimiento
Ambiente electromagnético: orientación
Prueba de
Nivel de prueba
inmunidad
IEC 60601
Radiofrecuencia
10 Vrms
10 Vrms
Los equipos portátiles y móviles deconducidacomunicaciones por radiofrecuencia no
150 kHz a 80 MHz
deben usarse más cerca de las piezas delIEC 61000-4-6
FreeStyle Comfort, incluidos los cables, quela distancia de separación recomendadacalculada a partir de la ecuación aplicable
10 V/m
10 V/ma la frecuencia del transmisor.
Radiofrecuencia
de 80 MHz a 2700 MHz
irradiada
IEC 61000-4-3

Distancia recomendada de separaciónd = 1.2 P

d = 1.2 P = 80 MHz a 800 MHz
d = 1.2 P = 800 MHz a 2700 MHz
donde P es la clasificación máxima depotencia de salida del transmisor en vatios(W) de acuerdo con el fabricante deltransmisor y d es la distancia de separación
recomendada en metros (m).
Las concentraciones de campo detransmisores fijos de radiofrecuencia,
determinadas por un relevamiento
electromagnético del sitio,a deberían ser
inferiores al nivel de cumplimiento en cadarango de frecuencia. b
Puede producirse interferencia en lacercanía de los equipos marcados con
el símbolo siguiente:

NOTA 1: A 80 MHz y 800 MHz, se aplica el rango de mayor frecuencia.
NOTA 2: Estas pautas pueden no ser de aplicación en todas las situaciones. La propagación electromagnética se ve
afectada por la absorción y la reflexión de estructuras, objetos y personas.
a
Las concentraciones de campo de los transmisores fijos, tales como las estaciones de base para teléfonos de radio(móviles/inalámbricos) y radios móviles terrestres, radios de aficionados y las transmisiones de radio de AM y FM y latransmisión de TV, no pueden predecirse teóricamente con precisión. Para evaluar el ambiente electromagnético debido
a transmisores fijos de radiofrecuencia, debe considerarse un estudio electromagnético del sitio. Si la concentración decampo medida en la ubicación donde se usa el FreeStyle Comfort supera el nivel correspondiente de cumplimiento deradiofrecuencia anterior, el FreeStyle Comfort debe observarse para verificar su funcionamiento normal. Si se observa unrendimiento anormal, pueden necesitarse medidas adicionales, tales como la reorientación o la reubicación del sistema
FreeStyle Comfort.
b

En el rango de frecuencia de 150 kHz a 80 MHz, las concentraciones de campo deben ser inferiores a 10 V/m.

136 - ES PN MN232-C4 L | Manual del usuario

FreeStyle Comfort
Distancias recomendadas de separación entre equipos portátiles y móviles
de comunicaciones de radiofrecuencia y el FreeStyle Comfort

El FreeStyle Comfort está destinado para su uso en un ambiente electromagnético en el que se controlan las interferencias
de radiofrecuencia irradiada. El cliente o el usuario del sistema FreeStyle Comfort puede ayudar a prevenir la interferencia
electromagnética al mantener una distancia mínima entre los equipos portátiles y móviles de comunicación por
radiofrecuencia (transmisores) y el FreeStyle Comfort, tal como se recomienda a continuación, de acuerdo
con la potencia máxima de salida de los equipos de comunicaciones.
Potencia de salida
Distancia de separación de acuerdo con la frecuencia del transmisor
nominal máxima delmtransmisor
150 kHz a 80 MHz
80 MHz a 800 MHz
800 MHz a 2,5 GHz
d = 1.2 P
d = 1.2 P
d = 1.2 P
W
0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
Para transmisores con una potencia máxima de salida no mencionada anteriormente, la distancia (d) de separación
recomendada en metros (m) puede calcularse usando la ecuación aplicable a la frecuencia del transmisor, donde P es lapotencia nominal máxima de salida del transmisor en vatios (W) de acuerdo con el fabricante del transmisor.
NOTA 1: A 80 MHz y 800 MHz, se aplica la distancia de separación para el rango de mayor frecuencia.
NOTA 2: Estas pautas pueden no ser de aplicación en todas las situaciones. La propagación electromagnética se ve
afectada por la absorción y la reflexión de estructuras, objetos y personas.

Clasificación

Tipo de protección frente a descargas eléctricas:
Clase II	La protección contra descargas eléctricas
			se consigue mediante doble aislamiento.

• CISPR 11 / EN 55011 Clase B Grupo 1,
	«Industrial, Scientific, and Medical (ISM)	Equipment»• FCC Parte 15, Subparte B-Clase B
	Radiadores involuntarios

Grado de protección frente a descargas eléctricas:
	Tipo BF Equipo que proporciona un		grado de protección frente a descargas		eléctricas relativo a		1) corriente de fuga permitida;		2) fiabilidad de la puesta a tierra de seguridad		(si procede).
		No está destinado para aplicación cardíaca directa.

IP22 - Protección contra la entrada de dedos u objetos
similares de diámetro superior a 12,5 mm. El goteo
vertical de agua no tendrá un impacto negativo si
la carcasa está inclinada en un ángulo de 15º de su
posición normal.

Pruebas independientes de estándares de equipos
eléctricos médicos:
	Verificado para cumplir con el reglamento:• I EC 60601-1 edición 3.1: equipos médicos
eléctricos, Parte 1: requisitos generales parala seguridad.
• CAN/CSA C22.2 n.º 60601-1-8 edición 2.1,
equipos médicos eléctricos, Parte 1: requisitos
generales para la seguridad.

Método permitido de limpieza y control de infección:
Consulte la sección «Limpieza, cuidados ymantenimiento adecuado» de este manual delusuario de FreeStyle.
Grado de seguridad de la aplicación en presencia
de gases anestésicos inflamables:	el equipo no está preparado para tal aplicación.
Modo de funcionamiento: funcionamiento continuo.

Protección contra interferencias electromagnéticas
potenciales o de otra índole entre el equipo y otros
dispositivos.
	• Probado para cumplir con el reglamento
EN 60601-1-2 /IEC 60601-1-2: 4ª edición.
	• Verificado para cumplir con el reglamento
RTCA/DO160 Sección 21 Categoría M.
PN MN232-C4 L | Manual del usuario ES - 137

FreeStyle Comfort
Controlli utente e indicatori di stato del sistema
ISO 7000
Manuale d'uso; istruzioni difunzionamento. N. reg. 1641

ISO 7010: Simboli grafici – Colori di sicurezza e
simboli di sicurezza – Simboli di sicurezza registrati
Leggere il manuale di istruzioni.
N. reg. M002

Paese e data di fabbricazione. "CC"identifica il codice a due lettere del Paesedi fabbricazione. La data di fabbricazione èin formato AAAA-MM-GG. N. reg. 6049

Conservare lontano da fiamme libere,
fuoco e scintille. È vietato fumare eutilizzare fonti di accensione aperte.
N. reg. P003

Tenere lontano dalla pioggia e
conservare in luogo asciutto. N. reg.
0626

Non fumare nei pressi dell'unità odurante il suo utilizzo. N. reg. P002

Limite di impilamento per numero, dove"n" indica il numero massimo consentito
di unità. N. reg. 2403
Numero di catalogo. N. reg. 2493

Avvertenza. N. reg. W001
IEC 60417
Parte applicata di tipo BF (gradodi protezione da scosse elettriche).
N. reg. 5333

Numero di serie. N. reg. 2498
Intervallo di temperatura diconservazione o di esercizio:
N. reg. 0632
Intervallo di umidità di conservazione.
N. reg. 2620
Limite della pressione atmosferica.
N. reg. 2621
Nome e indirizzo del produttore.
N. reg. 3082

Apparecchiatura di Classe II, doppio
isolamento N. reg. 5172
Regolamento UE 2017/745
Rappresentante autorizzato per la
Comunità Europea
Se l'etichetta di identificazione univocadel prodotto (UDI) riporta il simbolo
CE####, il dispositivo è conforme ai
requisiti relativi ai dispositivi medici.
Il simbolo CE#### indica il numero
dell'organismo notificato.
Simboli aggiuntivi

Lato alto. N. reg. 0623

Tenere lontano da materiali
infiammabili, olio e grasso.

Fragile, maneggiare con cura.
N. reg. 0621

Non smontare.

Contiene sostanze pericolose.
N. reg. 3723

Incremento del flusso.

Importatore. N. reg. 3725

Decremento del flusso.

Uscita O2

Pulsante Menu.
Interruttore di alimentazione. N. reg.
5009
Conforme a
RTCA/DO160 Sezione
21 CategoriaM

138 - IT PN MN232-C4 L | Manuale dell’utente

Il produttore di questo concentratore
di ossigeno portatile ha determinato che
il dispositivo è conforme a tutti i criteri
di accettazione FAA applicabili e all'uso
a bordo di aeromobili.

FreeStyle Comfort
Il produttore di questo concentratore
di ossigeno portatile ha determinato che
il dispositivo è conforme a tutti i criteri
di accettazione FAA applicabili e all'uso
a bordo di aeromobili.

Direttiva del Consiglio 2012/19/UE
RAEE
Questo simbolo ha lo scopo di ricordare
ai proprietari dell'apparecchiatura
di consegnarlo presso un centro peril riciclaggio al termine della durata
utile, in conformità alla Direttiva sui
rifiuti di apparecchiature elettriche ed
elettroniche (RAEE). I nostri prodottisono conformi alle restrizioni dettate
dalla Direttiva sulle sostanze pericolose(RoHS). È possibile che contengano
tracce, ma non quantità rilevanti, dipiombo o altri materiali pericolosi.

Riciclabile
Rappresentante autorizzato in Svizzera(ADD)
Se il dispositivo reca il marchio UKCA
come mostrato con UKCA####
indicante il numero di organismo
notificato, questo dispositivo è conforme
alle normative UKCA.
21 CFR 801.15
La legge federale degli Stati Uniti limita
la vendita del presente dispositivo ai soli
medici o dietro prescrizione medica.
Certificazione ETL

IEC 60601-1-8
Allarme tecnico a bassa priorità
ISO 15223-1

Certificato per i mercati statunitensee canadese in conformità ai relativi
standard applicabili.
Contiene ID FCC: WAP2001
Avviso FCC
Questo dispositivo può contenere
CYBLE-022001-00, ivi compresa l'antenna
2450AT18B100 di Johnson Technology, edè conforme alla parte 15 delle norme FCC.
Il dispositivo soddisfa i requisiti di approvazione deltrasmettitore modulare, come specificato nell'Avviso
pubblico FCC DA00-1407. Il funzionamentodel trasmettitore è soggetto alle seguenti due
condizioni: (1) questo dispositivo non può provocare
interferenze pericolose e (2) questo dispositivo
deve accettare eventuali interferenze ricevute, ivi
comprese interferenze che possono provocare un
funzionamento indesiderato. Il prodotto può essere
coperto da uno o più brevetti, validi negli Stati Uniti
e in altri Paesi. Visitare il nostro sito Web, sottoindicato, per l'elenco dei brevetti applicabili. Brev.:www.caireinc.com/corporate/patents
Simboli grafici per l'utilizzo sull'apparecchiatura
IEC 60417

Dispositivo medico. N. reg. 5.7.7
Codice identificativo unico deldispositivo 5.7.10
A S/NZS 4417.1, AS/NZS 4417.2
Marchio di conformità normativa
ACMA – indica la conformità ai
regolamenti ACMA (ADD)

Tecnologia wirelessTipi di connessione

SSP, iAP2, GATT

Frequenzada 2402 a 2480 MHz

Uscita massima dipotenza RF+4 dBmIntervallo di esercizio

10 m (classe 2)

È consigliabile collocare la macchina in funzionea una distanza minima di 1,1 cm (0,43 pollici) dal
corpo. Non applicabile a maschere, tubature per aria
o accessori.

Corrente continua N. reg. 5031
Corrente alternata N. reg. 5032
IEC 60601-1
A prova di gocciolamento

PN MN232-C4 L | Manuale dell’utente IT - 139

FreeStyle Comfort
Estrazione del sistema FreeStyle Comfort dalla confezione
Verificare che la confezione
contenga tutti i componenti
elencati di seguito. In caso dielementi mancanti, contattare
immediatamente il fornitore diossigeno.

Accessori standard• Concentratore di ossigeno
portatile• Borsa per il trasporto• Tracolla• Gruppo batterie a 8 celle• Alimentatore CA• Cavo di alimentazione in CA• Alimentatore CC• Manuale dell'utente

Accessori opzionali• Gruppo batterie a 8 celle
aggiuntivo• Gruppo batterie a 16 celle• Caricatore desktop• Borsa per il trasporto degli
accessori

Cenni preliminari sul sistema FreeStyle Comfort
Acquisire familiarità con le funzionalità chiave del sistema FreeStyle Comfort e l'interfaccia utente.

Vista frontale
Attacco della tracolla

Interfaccia utente

Uscitadell'ossigeno
Uscita dell'aria

Ingressodell'aria/Filtri
Ingresso dell'aria/
Filtri
Linguetta dellapresa della porta
di alimentazione
Gruppo batterie
Pulsante di sganciodel gruppo batterie
140 - IT PN MN232-C4 L | Manuale dell’utenteFreeStyle Comfort
Vista dall'alto
Display
Incrementodel flusso

Pulsante diaccensione

Pulsante Menu

Decrementodel flusso

Display della batteria
Impostazionedel flusso

Display dellabatteria

Indicatore autoSAT
Percentuale
di carica
rimanente

Visualizzazione
allarmiIndicatore
somministrazione
bolo

Display dell'alimentatore CA/CC

Impostazionedel flusso

Area divisualizzazione
degli allarmi

Display dellabatteria

Indicatoredi ricarica

PN MN232-C4 L | Manuale dell’utente IT - 141

FreeStyle Comfort
Cosa si intende per concentratore
di ossigeno
L'aria che respiriamo è composta all'incirca dal 21%
di ossigeno, 78% di azoto e 1% di altri gas. Nelsistema FreeStyle® Comfort®, l'aria dell'ambiente viene
aspirata nel dispositivo attraverso gli ingressi dell'aria,
passando poi attraverso un materiale assorbente
denominato setaccio molecolare. Questo materiale
separa ossigeno e azoto, consentendo il passaggio solodell'ossigeno. Il risultato è l'erogazione all'utilizzatore
di un flusso di ossigeno ad alta concentrazione.
Il sistema FreeStyle Comfort combina le tecnologieavanzate dei concentratori di ossigeno e diconservazione dell'ossigeno per creare un dispositivo
portatile che permette la massima portabilità e ladeambulazione. L'erogazione avanzata con flusso a
impulsi rileva rapidamente quando l'utente inspira ed
eroga un impulso di ossigeno all'inizio di ciascunainalazione. L'erogazione con flusso a impulsi è ideale
per le attività e i periodi fuori casa, conserva l'ossigeno
ed estende la durata della batteria rispetto al flusso
continuo. Il sistema FreeStyle Comfort è abbastanza
leggero da poter essere trasportato con una tracolla.
Può funzionare con CA (rete domestica), CC (veicolo)o a batteria.

Uso previsto
Il concentratore di ossigeno CAIRE FreeStyle
Comfort è destinato alla somministrazione di ossigeno
supplementare. Il dispositivo non è progettato per il
supporto vitale artificiale e non offre funzionalità dimonitoraggio del paziente.

Perché il medico prescrive
l'ossigenoterapia?
Molte persone soffrono di svariate patologie cardiache,polmonari e respiratorie di altro tipo. Un numero
significativo di queste persone può trarre beneficio da
un'ossigenoterapia supplementare a casa, in viaggio odurante la partecipazione ad attività quotidiane lontanoda casa.
L'ossigeno è un gas che compone il 21% dell'aria che
respiriamo; per funzionare correttamente, l'organismo
ne richiede una fornitura costante. Il medico prescrive
un flusso o un'impostazione per fronteggiare una
condizione respiratoria particolare.

142 - IT PN MN232-C4 L | Manuale dell’utente

Benché l'ossigeno sia un farmaco che non produce
dipendenza, l'ossigenoterapia non autorizzata puòessere pericolosa. È necessario consultare un medico
prima di usare questo concentratore di ossigeno.
Il fornitore dell'apparecchiatura per l'ossigeno
dimostrerà come impostare la portata prescritta.

Profilo operatore
I concentratori sono destinati a fornire ossigeno
supplementare agli utilizzatori che soffrono di disagi
causati da disturbi che incidono sull'efficienza dei
polmoni nel trasferimento dell'ossigeno nell'aria alsangue. I concentratori di ossigeno portatili (COP)
non conservano né contengono ossigeno. Non devono
essere riforniti e possono essere ricaricati ovunquesia disponibile alimentazione CA o CC (tranne chea bordo di un aeromobile). In questo modo l'utenteè relativamente autosufficiente in termini di utilizzo in
casa, deambulazione (sia all'interno sia all'esterno dellacasa), mobilità e stile di vita in generale. L'utilizzodel concentratore di ossigeno richiede una prescrizionedel medico.
Il tipico utente dell'ossigenoterapia ha superato
i 65 anni e soffre di svariate malattie respiratorie, tracui la broncopneumopatia cronica ostruttiva (BPCO).
Gli utenti in genere dispongono di buone capacità
cognitive e devono essere in grado di comunicare
il disagio. Se l'utente non è in grado di comunicare
il disagio o di leggere e comprendere le etichette e
le istruzioni per l'uso del concentratore, l'utilizzo è
consigliato solo sotto supervisione. In caso di disagio
durante l'uso del concentratore, gli utenti sono invitati
a rivolgersi al proprio fornitore di assistenza sanitaria.
Gli utenti sono inoltre invitati a disporre di una fonte diossigeno di riserva (ovvero, una bombola di ossigeno)
in caso di interruzione dell'alimentazione o guasto delconcentratore. Non sono previste altre abilità peculiario necessarie dell'utente per l'uso del concentratore.

FreeStyle Comfort
Importante!
Le istruzioni per la sicurezza sono definite comeindicato di seguito:
AVVERTENZA: IMPORTANTI INFORMAZIONI PER LA SICUREZZA RELATIVEAI PERICOLI CHE POSSONO CAUSAREGRAVI LESIONI.
ATTENZIONE: informazioni importanti
per prevenire danni al sistema FreeStyle
Comfort.
Nota: informazioni che richiedono particolare attenzione.

Indicazioni per l'uso
AVVERTENZA: È MOLTO IMPORTANTESELEZIONARE SOLO IL LIVELLO DI OSSIGENO PRESCRITTO. NON MODIFICARELA SELEZIONE DEL FLUSSO SE NON IN
CASO DI PRECISE ISTRUZIONI DA PARTE
DI UN MEDICO AUTORIZZATO.
AVVERTENZA: QUESTO DISPOSITIVO NON È STATOTESTATO NÉ APPROVATO PER L'USO IN PAZIENTIPEDIATRICI.
AVVERTENZA: LA LEGGE FEDERALE (USA) LIMITALA VENDITA O IL NOLEGGIO DI QUESTO DISPOSITIVODIETRO PRESCRIZIONE DI UN MEDICO O DI UN
ALTRO PROFESSIONISTA SANITARIO QUALIFICATO.
AVVERTENZA: QUESTA UNITÀ NON DEVE ESSERE
USATA COME SUPPORTO SALVAVITA. POTREBBEESSERE NECESSARIO UN MONITORAGGIOAGGIUNTIVO PER I PAZIENTI NON IN GRADO DI
COMUNICARE I PROPRI DISTURBI DURANTE L'USO
DEL DISPOSITIVO. GLI UTENTI CON DANNI ALL'UDITO
E/O ALLA VISTA POTREBBERO NECESSITARE DIASSISTENZA PER MONITORARE GLI ALLARMI.
AVVERTENZA: LE IMPOSTAZIONI DELLA MODALITÀDOSAGGIO IMPULSI DEVONO ESSERE DEFINITE PERCIASCUN UTENTE INDIVIDUALMENTE, IN FUNZIONEDELLE RELATIVE ESIGENZE A RIPOSO, DURANTE
L'ESERCIZIO FISICO E DURANTE I VIAGGI.
AVVERTENZA: QUESTO DISPOSITIVO È DESTINATO
ALL'USO NELL'ASSISTENZA SANITARIADOMICILIARE E IN AMBIENTI AMBULATORIALI SIMILI.

AVVERTENZA: ALCUNI SFORZI RESPIRATORIDEL PAZIENTE POTREBBERO NON ATTIVARE LAFUNZIONE DI CONSERVAZIONE.
AVVERTENZA: IL SISTEMA FREESTYLE COMFORTNON È ADATTO ALL'USO CON UN PAZIENTE
TRACHEOTOMIZZATO.
AVVERTENZA: LE DONNE INCINTE O CHE
ALLATTANO NON DEVONO USARE GLI ACCESSORIRACCOMANDATI IN QUESTO MANUALE, POICHÉPOSSONO CONTENERE FTALATI.

Controindicazioni all'uso
AVVERTENZA: IN ALCUNE CIRCOSTANZE,L'USO DI OSSIGENO NON PRESCRITTO
PUÒ ESSERE PERICOLOSO. QUESTODISPOSITIVO DEVE ESSERE USATO SOLODIETRO PRESCRIZIONE MEDICA.
AVVERTENZA: QUESTO DISPOSITIVO NON ÈDESTINATO ALL'USO ALL'INTERNO O IN PROSSIMITÀ
DI AMBIENTI DI RISONANZA MAGNETICA, COME
AD ESEMPIO MACCHINE PER RISONANZA
MAGNETICA (RM).
AVVERTENZA: NON USARE IN PRESENZADI ANESTETICI INFIAMMABILI.
AVVERTENZA: ANALOGAMENTE A QUALSIASIALTRO DISPOSITIVO ELETTRICO, POTREBBERO
VERIFICARSI PERIODI DI INATTIVITÀ DOVUTIALL'INTERRUZIONE DELL'ALIMENTAZIONEELETTRICA, OPPURE ALLA NECESSITÀ DI FARRIPARARE IL CONCENTRATORE DI OSSIGENO DA
UN TECNICO QUALIFICATO. IL CONCENTRATOREDI OSSIGENO NON È INDICATO PER UTENTI CHE
A SEGUITO DELL'INTERRUZIONE TEMPORANEAPOTREBBERO SUBIRE CONSEGUENZE NEGATIVEPER LA PROPRIA SALUTE.
AVVERTENZA: QUESTO DISPOSITIVO NONÈ APPROVATO PER L'USO IN TUTTI GLIAMBIENTI. L'ESPOSIZIONE AD ALCUNEFONTI ELETTROMAGNETICHE PUÒ CAUSARE
INTERFERENZE O PRESTAZIONI INVOLONTARIEDEL DISPOSITIVO. IN CASO DI INTERFERENZE,
SPEGNERE IMMEDIATAMENTE IL DISPOSITIVO.

AVVERTENZA: QUESTO DISPOSITIVO È DESTINATO
ALL'USO DA PARTE DEL PAZIENTE CHE RICEVEL'OSSIGENOTERAPIA.

PN MN232-C4 L | Manuale dell’utente IT - 143

FreeStyle Comfort
Linee guida per la sicurezza
AVVERTENZA: NON È CONSENTITOAPPORTARE ALCUNA MODIFICA ALDISPOSITIVO
AVVERTENZA: QUESTO DISPOSITIVO EROGAOSSIGENO A ELEVATA CONCENTRAZIONE CHE
FAVORISCE IL RAPIDO SVILUPPO DI USTIONI. NON
CONSENTIRE AL PAZIENTE DI FUMARE O DI USARE
FIAMME LIBERE ALL'INTERNO DELLA STANZAIN CUI SI TROVA (1) QUESTO DISPOSITIVO O (2)QUALSIASI ALTRO ACCESSORIO PER IL TRASPORTO
DELL'OSSIGENO. IL MANCATO RISPETTO DI QUESTA
AVVERTENZA PUÒ COMPORTARE GRAVI INCENDI EDANNI MATERIALI E/O PROVOCARE LESIONI FISICHEO IL DECESSO.
AVVERTENZA: NON USARE L'UNITÀ IN SPAZICONFINATI O ANGUSTI (AD ESEMPIO UNA BORSA OUNA CUSTODIA DI PICCOLE DIMENSIONI) IN CUI LAVENTILAZIONE POSSA ESSERE LIMITATA. IN QUESTO
MODO IL CONCENTRATORE DI OSSIGENO POTREBBESURRISCALDARSI PREGIUDICANDO LE PRESTAZIONI.
AVVERTENZA: IL CONCENTRATORE VA POSIZIONATO
IN MODO DA EVITARE FUMO, AGENTI INQUINANTIO VAPORI.
AVVERTENZA: NEL CASO IN CUI SI VERIFICHIUN GRAVE INCIDENTE CON QUESTO DISPOSITIVO,
L'UTENTE DEVE IMMEDIATAMENTE SEGNALAREL'INCIDENTE AL FORNITORE E/O AL PRODUTTORE.
UN INCIDENTE GRAVE È DEFINITO COME UNALESIONE, UN DECESSO O UNA POTENZIALE CAUSADI LESIONE/MORTE NEL CASO IN CUI SI VERIFICHINUOVAMENTE L'INCIDENTE. L'UTENTE PUÒ
ANCHE SEGNALARE L'INCIDENTE ALL'AUTORITÀ
COMPETENTE DEL PAESE IN CUI SI È VERIFICATO.

AVVERTENZA: L'USO DI ALCUNIACCESSORI PER LA SOMMINISTRAZIONEDI OSSIGENO NON ESPRESSAMENTE
INDICATI PER L'USO CON QUESTOCONCENTRATORE DI OSSIGENO PUÒ
PREGIUDICARNE LE PRESTAZIONI.
GLI ACCESSORI RACCOMANDATISONO RIPORTATI ALL'INTERNO
DEL PRESENTE MANUALE.
AVVERTENZA: IN CASO DI CADUTA, DANNO OESPOSIZIONE ALL'ACQUA DEL CONCENTRATOREDI OSSIGENO, RIVOLGERSI AL FORNITORE
DELL'APPARECCHIATURA PER UN'ISPEZIONE O
UN'EVENTUALE RIPARAZIONE DEL DISPOSITIVO.
NON USARE IL CONCENTRATORE DI OSSIGENOSE IL CAVO O LA SPINA DI ALIMENTAZIONE SONODANNEGGIATI.
AVVERTENZA: DURANTE IL TRASPORTO
ALL'INTERNO DI UN VEICOLO, FISSARE IL
CONCENTRATORE DI OSSIGENO IN MANIERAADEGUATA, MEDIANTE UNA CINTURA O UN FERMODI ALTRO TIPO, PER EVITARE DANNI O LESIONI.
144 - IT PN MN232-C4 L | Manuale dell’utente

AVVERTENZA: IL DEPARTMENT OF TRANSPORTATION(DOT) DEGLI STATI UNITI E I REGOLAMENTI DELLE
NAZIONI UNITE RICHIEDONO LA RIMOZIONEDELLA BATTERIA DAL DISPOSITIVO PER L'INTERO
VIAGGIO AEREO INTERNAZIONALE QUANDO IL
CONCENTRATORE DI OSSIGENO VIENE REGISTRATO
COME BAGAGLIO. DURANTE LA SPEDIZIONE DEL
CONCENTRATORE DI OSSIGENO, È NECESSARIORIMUOVERE LA BATTERIA DAL DISPOSITIVO E
IMBALLARLO IN MANIERA ADEGUATA.
AVVERTENZA: NON USARE QUESTO DISPOSITIVOSENZA PRIMA AVER LETTO E COMPRESO IL
PRESENTE MANUALE. IN CASO DI DIFFICOLTÀ
A COMPRENDERE LE AVVERTENZE E LEISTRUZIONI, RIVOLGERSI AL FORNITORE
DELL'APPARECCHIATURA PRIMA DI TENTAREDI USARLA; IN CASO CONTRARIO, POSSONOVERIFICARSI DANNI O LESIONI.
AVVERTENZA: SE SI MANIFESTANO DISTURBIO IN CASO DI EMERGENZA MEDICA, RIVOLGERSIIMMEDIATAMENTE A UN MEDICO.
AVVERTENZA: L'UTILIZZO DEL CONCENTRATOREDI OSSIGENO AL DI FUORI DELLE SPECIFICHE DITEMPERATURA DI ESERCIZIO PUÒ LIMITARE LACAPACITÀ DEL CONCENTRATORE DI SODDISFAREI VALORI SPECIFICATI PER LA CONCENTRAZIONE
DELL'OSSIGENO. PER I LIMITI DI TEMPERATURA,
FARE RIFERIMENTO ALLA SEZIONE DELLE
SPECIFICHE RIPORTATA ALL'INTERNO DELPRESENTE MANUALE.
AVVERTENZA: NON USARE PRODOTTI A
BASE DI OLIO, GRASSO O PETROLIO O ALTRIPRODOTTI INFIAMMABILI CON GLI ACCESSORIPER IL TRASPORTO DELL'OSSIGENO O IL
CONCENTRATORE DI OSSIGENO. USAREESCLUSIVAMENTE LOZIONI O UNGUENTI A
BASE D'ACQUA COMPATIBILI CON L'OSSIGENO.
L'OSSIGENO ACCELERA LA COMBUSTIONE DELLESOSTANZE INFIAMMABILI.
AVVERTENZA: L'USO NON CORRETTO DELLABATTERIA PUÒ CAUSARNE IL SURRISCALDAMENTO
E L'ACCENSIONE, PROVOCANDO GRAVI LESIONI.
PRESTARE ATTENZIONE A NON BUCARE, COLPIRE,SALIRE SOPRA LA BATTERIA O FARLA CADERE,
E NON ESPORLA A IMPATTI O SOLLECITAZIONIFORTI. L'USO DI UNA BATTERIA DANNEGGIATA PUÒ
CAUSARE LESIONI PERSONALI.
AVVERTENZA: NON CORTOCIRCUITARE I CONTATTIIN METALLO DELLA BATTERIA CON OGGETTIMETALLICI, COME CHIAVI O MONETE; POTREBBERO
PRODURSI SCINTILLE O UN CALORE ECCESSIVO.
AVVERTENZA: DURANTE L'UTILIZZO DELCONCENTRATORE DI OSSIGENO PORTATILE
ALL'APERTO CON L'ALIMENTATORE CA,
COLLEGARE L'ALIMENTATORE SOLO A UNA PRESACON INTERRUTTORE DIFFERENZIALE (GFI).

FreeStyle Comfort
AVVERTENZA: NON USARE IL CONCENTRATOREDI OSSIGENO ACCANTO O IMPILATO SU ALTRE
APPARECCHIATURE. SE NON È POSSIBILE EVITARETALE MODALITÀ DI UTILIZZO, OSSERVARE ILDISPOSITIVO PER VERIFICARNE IL NORMALEFUNZIONAMENTO.
AVVERTENZA: L'USO DI CAVI E ADATTATORI DIVERSIDA QUELLI SPECIFICATI, A ECCEZIONE DI QUELLIVENDUTI DAL PRODUTTORE DEL DISPOSITIVOELETTROMEDICALE COME PARTI DI RICAMBIOPER I COMPONENTI INTERNI, PUÒ PROVOCAREUN AUMENTO DELLE EMISSIONI E UNA RIDUZIONE
DELL'IMMUNITÀ DEL CONCENTRATORE DI OSSIGENO.
AVVERTENZA: USARE SOLO LA TENSIONEELETTRICA SPECIFICATA SULL'ETICHETTA DELLE
SPECIFICHE APPLICATA SUL DISPOSITIVO.
AVVERTENZA: NON USARE PROLUNGHE CON
QUESTA UNITÀ O EVITARE DI COLLEGARETROPPE SPINE ALLA STESSA PRESA ELETTRICA.
L'USO DI PROLUNGHE PUÒ AVERE EFFETTI
AVVERSI SULLE PRESTAZIONI DEL DISPOSITIVO.
IL COLLEGAMENTO DI TROPPE SPINE A UN'UNICAPRESA PUÒ PROVOCARE UN SOVRACCARICO DELQUADRO ELETTRICO, CAUSANDO L'ATTIVAZIONE
DELL'INTERRUTTORE/FUSIBILE O UN INCENDIOIN CASO DI MANCATO FUNZIONAMENTO
DELL'INTERRUTTORE/FUSIBILE.
ATTENZIONE: in caso di allarme o qualora
si osservi un funzionamento non correttodel concentratore di ossigeno, consultare
la sezione Risoluzione dei problemi delpresente manuale. Se non si riesce arisolvere il problema, rivolgersi al fornitore
dell'apparecchiatura.
ATTENZIONE: per evitare l'annullamento dellagaranzia, attenersi a tutte le istruzioni del produttore.

AVVERTENZA: NON TENTAREDI ESEGUIRE INTERVENTI DI
MANUTENZIONE, A ECCEZIONE DELLE
SOLUZIONI POSSIBILI ELENCATE
ALL'INTERNO DEL PRESENTE MANUALE.
AVVERTENZA: IMPIEGARE ESCLUSIVAMENTEACCESSORI RACCOMANDATI DAL PRODUTTORE.
L'USO DI ALTRI ACCESSORI PUÒ ESSEREPERICOLOSO, CAUSARE GRAVI DANNI AL
CONCENTRATORE DI OSSIGENO E RENDERE NULLALA GARANZIA.
AVVERTENZA: IL PRODUTTORE CONSIGLIA DI
TENERE A DISPOSIZIONE UN'ALTRA SORGENTE
DI OSSIGENO SUPPLEMENTARE IN CASO DI
INTERRUZIONI DI CORRENTE, CONDIZIONI DI
ALLARME O GUASTI MECCANICI. PER CONOSCEREIL TIPO DI SISTEMA DI RISERVA NECESSARIO,RIVOLGERSI AL PROPRIO MEDICO O AL FORNITORE
DELL'APPARECCHIATURA.
AVVERTENZA: NON FAR BLOCCARE L'INGRESSOO GLI SFIATI DI USCITA DELL'ARIA. NON FARCADERE O INSERIRE ALCUN OGGETTO NELLE
APERTURE PRESENTI SUL DISPOSITIVO. IN QUESTO
MODO IL CONCENTRATORE DI OSSIGENO
POTREBBE SURRISCALDARSI PREGIUDICANDO LE
PRESTAZIONI.
AVVERTENZA: FUMARE DURANTE L'USO
DELL'OSSIGENO COSTITUISCE LA CAUSA NUMERO
UNO DI USTIONI E MORTI CORRELATE AGLI INCENDI.
È NECESSARIO ATTENERSI ALLE SEGUENTIAVVERTENZE DI SICUREZZA:
AVVERTENZA: NON CONSENTIRE AL PAZIENTEDI FUMARE O DI USARE CANDELE O FIAMME
LIBERE ALL'INTERNO DELLA STANZA IN CUI SI
TROVA IL DISPOSITIVO OPPURE QUALSIASI ALTROACCESSORIO PER L'EROGAZIONE DELL'OSSIGENO.
AVVERTENZA: FUMARE MENTRE SI INDOSSA UNACANNULA PER L'OSSIGENO PUÒ PROVOCAREUSTIONI AL VISO E PROBABILMENTE LA MORTE.
AVVERTENZA: SE SI FUMA, QUESTI 3 PASSAGGIPOSSONO SALVARE LA VITA: SPEGNERE IL
CONCENTRATORE DI OSSIGENO, RIMUOVERE LACANNULA E LASCIARE LA STANZA IN CUI SI TROVAIL DISPOSITIVO.

PN MN232-C4 L | Manuale dell’utente IT - 145

FreeStyle Comfort
AVVERTENZA: IN CASA O NEL LUOGO IN CUI SI
UTILIZZA L'OSSIGENO OCCORRE AFFIGGERECARTELLI "NON FUMARE – OSSIGENO IN USO"
IN MODO CHE SIANO BEN VISIBILI. GLI UTENTI ECOLORO CHE PRESTANO ASSISTENZA DEVONOESSERE INFORMATI DEI PERICOLI DEL FUMO INPRESENZA O DURANTE L'UTILIZZO DI OSSIGENO
MEDICO.
AVVERTENZA: NON USARE IL CONCENTRATOREDI OSSIGENO IN PRESENZA DI GAS INFIAMMABILI,POICHÉ POSSONO VERIFICARSI FENOMENIDI COMBUSTIONE RAPIDA, CHE POSSONOCOMPORTARE DANNI MATERIALI, LESIONICORPOREE O IL DECESSO.
AVVERTENZA: RIMUOVENDO LA CANNULA E
POSIZIONANDOLA SU INDUMENTI, BIANCHERIA,DIVANI O ALTRI MATERIALI IMBOTTITI SI POSSONOVERIFICARE LAMPI DI FUOCO IN CASO DIESPOSIZIONE A SIGARETTE, FONTI DI CALOREO FIAMME.
AVVERTENZA: NON LASCIARE LA CANNULA NASALE
SOPRA O SOTTO INDUMENTI, COPERTE O CUSCINIDI SEDIE. SE L'UNITÀ È ACCESA MA NON IN USO,L'OSSIGENO PUÒ RENDERE INFIAMMABILE IL
MATERIALE. ASSICURARSI CHE L'ALIMENTAZIONESIA SPENTA QUANDO IL CONCENTRATOREDI OSSIGENO NON È IN USO.
AVVERTENZA: GLI ACCESSORI APPROVATI PERL'EROGAZIONE DI OSSIGENO SONO PROGETTATIPER RIDURRE LA PROPAGAZIONE DEL FUOCO PERLA SICUREZZA DEL PAZIENTE. L'USO DI ACCESSORINON APPROVATI PUÒ AUMENTARE IL RISCHIODI LESIONI O DANNI CAUSATI DA INCENDI.
AVVERTENZA: IL CONCENTRATORE DI OSSIGENO
DEVE ESSERE SPENTO E SCOLLEGATO
DALL'ALIMENTAZIONE ESTERNA PRIMA DI
RIMUOVERE IL GRUPPO BATTERIE.
AVVERTENZA: PICCOLE PARTIPOSSONO PRESENTARE UN PERICOLODI SOFFOCAMENTO.
AVVERTENZA: EVITARE IL CONTATTO CON LA
BOCCHETTA DI SCARICO PER PIÙ DI 10 SEC PERCHÉPOTREBBE ESSERE CALDA.
AVVERTENZA: VENTO O CORRENTI D'ARIA
FORTI POSSONO INFLUIRE NEGATIVAMENTE
SULL'EROGAZIONE DELL'OSSIGENOTERAPIA.
AVVERTENZA: IL PRESENTE PRODOTTO PUÒESPORRE L'UTENTE A SOSTANZE CHIMICHE,TRA CUI IL NICHEL, CHE, SECONDO LO STATODELLA CALIFORNIA, PUÒ CAUSARE IL CANCRO.
PER MAGGIORI INFORMAZIONI, VISITARE
WWW.P65WARNINGS.CA.GOV.
ATTENZIONE: scollegare sempre
l'alimentatore CA dalla presa di rete
prima di scollegarlo dal concentratore
di ossigeno.
146 - IT PN MN232-C4 L | Manuale dell’utente

ATTENZIONE: non posizionare il concentratore diossigeno in modo che il cavo di alimentazione sia
difficile da raggiungere.
ATTENZIONE: sistemare sempre i tubi di mandata
dell'ossigeno e i cavi di alimentazione in modo daprevenire pericoli di inciampo o lo strangolamento
accidentale.
ATTENZIONE: quando si utilizza il concentratore
di ossigeno in auto, in barca o con altre sorgenti
CC tramite l'alimentatore CC, accertarsi di avviare
e mettere in funzione il veicolo prima di collegare
il concentratore di ossigeno. La mancata osservanzadi tali istruzioni può comportare la mancata
alimentazione del concentratore di ossigeno da partedell'alimentatore.
ATTENZIONE: quando si spegne l'automobile in cui
si utilizza il concentratore di ossigeno, scollegare e
rimuovere il dispositivo dal veicolo. Non conservare
il concentratore di ossigeno in un'automobile
molto calda o molto fredda o in altri ambienti simili
a temperatura molto alta o bassa.
Nota: le apparecchiature mobili di comunicazione
a RF possono avere effetti sui dispositivi elettromedicali.
ATTENZIONE: NON lasciare il
concentratore di ossigeno e l'alimentatore
collegati al veicolo se il sistema diaccensione è spento. Si potrebbe
scaricare la batteria del veicolo.
ATTENZIONE: l'utilizzo o la conservazione delconcentratore di ossigeno al di fuori del normale
intervallo di temperatura operativa o di pressioneambientale (altitudine) può influire sulle prestazioni
e ridurre il tempo di funzionamento della batteria
e/o aumentarne il tempo di ricarica. Per i limiti ditemperatura di funzionamento e conservazione, fareriferimento alla sezione delle specifiche riportata
all'interno del presente manuale.
ATTENZIONE: per i concentratori di ossigeno dotati
di batterie: conservare in un luogo fresco e asciutto
per contribuire a garantire la longevità della batteria.
La conservazione del concentratore di ossigeno
per periodi prolungati a temperature elevate o conla batteria completamente carica o completamente
scarica può ridurre la durata complessiva dellabatteria. Non tentare di aprire la batteria, che noncontiene componenti riparabili. Tenere le batterie
lontano dai bambini.
ATTENZIONE: USARE ESCLUSIVAMENTE le batterie
fornite dal produttore. Per il corretto smaltimento dellabatteria, rivolgersi al fornitore dell'apparecchiaturao all'ente governativo preposto per i requisiti dismaltimento.

FreeStyle Comfort
Nota: per mitigare eventuali danni causati da un guastodel concentratore o dall'interruzione dell'alimentazione,
il fornitore dell'apparecchiatura deve valutare lanecessità di ossigeno di riserva durante l'installazionee periodicamente al variare dei seguenti fattori:

Nota: se il concentratore di ossigeno è stato conservato
per un periodo prolungato al di fuori delle normali
condizioni ambientali operative indicate nella tabella delle
specifiche, prima dell'accensione attendere che torni alla
normale temperatura di funzionamento.

- Le condizioni del paziente

Nota: il fornitore dell'apparecchiatura deve valutare lanecessità di ossigeno di riserva in caso di guasto delconcentratore o di interruzione dell'alimentazione durante
l'installazione e periodicamente al variare dei seguenti fattori:- L'ambiente in cui vive il paziente- La capacità di rifornire il paziente con ossigeno
supplementare di riserva.
Nota: accertarsi che la cannula sia completamente
inserita e salda. In questo modo il concentratore
di ossigeno è in grado di rilevare correttamente
l'inspirazione per erogare correttamente l'ossigeno.
Durante l'inalazione, si deve udire o avvertire il flusso diossigeno attraverso la biforcazione della cannula nasale.

- Le condizioni del paziente- L'ambiente in cui vive il paziente- La capacità di rifornire il paziente con ossigeno
supplementare di riserva.

Nota: le batterie al litio possono perdere in modo
permanente la capacità se esposte a temperature
estremamente elevate con le batterie completamente
cariche o completamente esaurite. Per la conservazione
prolungata, si raccomanda di caricare le batterie tra il
25 e il 50% e di conservarle a una temperatura di-20 °C – 20 °C (-4 °F – 68 °F).
Nota: non usare il concentratore di ossigeno senza il filtro
di ingresso dell'aria in posizione. Se è disponibile unsecondo filtro, inserire il filtro di ricambio prima di pulire
quello sporco. Pulire il filtro sporco con una soluzione
tiepida di acqua e sapone e asciugare a fondo primadell'uso.
Nota: il produttore sconsiglia la sterilizzazione di questo
dispositivo.
Nota: quando si usa l'unità del concentratore di ossigeno
non esiste mai il rischio di esaurire l'ossigeno presente inuna stanza.
AVVERTENZA: IL CONCENTRATOREDI OSSIGENO NON VA UTILIZZATOO COLLEGATO ALL'ALIMENTAZIONEESTERNA (CA/CC) SENZA LA BATTERIAINSTALLATA. NON TENTARE DI
ACCENDERE L'UNITÀ O COLLEGARLAALL'ALIMENTAZIONE ESTERNA QUANDOIL GRUPPO BATTERIE È RIMOSSO.
RIMUOVERE IL GRUPPO BATTERIE SOLO QUANDOL'APPARECCHIATURA È SPENTA E SCOLLEGATA
DALL'ALIMENTAZIONE. NON TENTARE DIRIMUOVERE IL GRUPPO BATTERIE MENTRE
L'UNITÀ È ANCORA ACCESA O COLLEGATA
ALL'ALIMENTAZIONE.

PN MN232-C4 L | Manuale dell’utente IT - 147

FreeStyle Comfort
Specifiche
FreeStyle Comfort
Dimensioni (con batteria singola)

25,4 x 18,5 x 7,9 cm (10.0 x 7.3 x 3.1 in)Dimensioni (con batteria doppia)

27,9 x 18,5 x 7,9 cm (11.0 x 7.3 x 3.1 in)

Peso (con gruppo batterie singolo)

2,3 kg (5 lb)

Peso (con gruppo batterie doppio)

2,7 kg (6 lb)

Livello di rumorosità nominale***39,93 dB(A) ± 0,42 dB(A) a 2

Impostazioni di flusso
Impostazione di uscita O2([a STPD, 20 bpm] Tutti i volumi +/- 15%)

Concentrazione di ossigeno*
Monitor dell'ossigeno
Pressione di uscita massima
Sensibilità di somministrazione

Impostazione di flusso 1, 2, 3, 4, 5
Impostazione 1 – 210 ml/min
Impostazione 2 – 420 ml/min
Impostazione 3 – 630 ml/min
Impostazione 4 – 840 ml/min
Impostazione 5 – 1050 ml/min
90% (+5,5% / -3%)
Sì
<30 PSI> -0,1 cm H2O (regolabile)

Alimentazione CA

100-240 V CA, 50-60 Hz, 120 W max

Alimentazione CC

11-18 V CC (10 A max)

Tipo di batteria
Capacità della batteria
Tempo di ricarica della batteria

Ioni di litio
Cella singola: 6700 mAh; Cella doppia: 13400 mAh
Batteria singola: 3,5 ore
Batteria doppia: 6,0 ore

Durata dei gruppi batterie

Durata batteria singola: Impostazione 2 – fino a 4 ore
Durata batteria doppia: Impostazione 2 – fino a 8 ore

Temperatura di esercizio**

5 °C - 40 °C (41 °F - 104 °F)

Temp. di trasporto e conservazione (unità)

Da -25 a +70 °C (da -13 a +158 °F), fino a 3048 m (10000 ft)

Temp. di trasporto e conservazione (batteria)

Da -25 a +70 °C (da -13 a +158 °F), fino a 3048 m (10000 ft)

Altitudine di esercizio**
Umidità di esercizio
Umidità di conservazione
Garanzia limitata

Da -381 a 3048 m (da -1250 a 10.000 ft) (testata a 700-1060 hPa)
15 - 95% di umidità relativa (senza condensa)
da 0% a 90% senza condensa
Vedere la dichiarazione di garanzia*Sulla base di una pressione atmosferica di 101 kPa (14,7 psi) a 21 °C (70 °F)**L'unità funzionerà in tutti gli ambienti che rientrano negli intervalli di temperatura e umidità specificati. Il
funzionamento al di fuori di queste specifiche operative può limitare la capacità del concentratore di soddisfare lespecifiche di concentrazione dell'ossigeno alle portate più elevate in litri. La schermata del riscaldamento può continuare
a comparire finché il concentratore non ha raggiunto la concentrazione desiderata o per un periodo massimo di 15 minuti.
Continuare a usare l'unità durante questo periodo.
***Livello sonoro misurato con il metodo di test n. 14-1 10/2018 MDS-Hi.
La durata prevista è di almeno cinque anni.

148 - IT PN MN232-C4 L | Manuale dell’utenteFreeStyle Comfort
Istruzioni per l'uso
1. Collocare e sistemare il sistema FreeStyle Comfort
in una posizione appropriata per l'uso.
a. Il concentratore deve essere ben ventilato inmodo che i filtri di ingresso dell'aria e l'uscita
dell'aria non siano ostruiti.
b. Il concentratore deve essere posizionato inmodo che tutti gli indicatori acustici e visivi
e gli allarmi possano essere facilmente visti esentiti.
c. Prima di usare l'unità, accertarsi che i filtri
di ingresso dell'aria siano in posizione. Se il
filtro dell'unità manca, contattare il fornitore
dell'apparecchiatura.
2. Installare un gruppo batterie nel concentratore.
a. Allineare la batteria con l'alloggiamento
inferiore dal centro del sistema FreeStyle
Comfort.

3. Inserire l'alimentazione CA o CC nella presa dellaporta di alimentazione sulla parte anteriore, in basso
a destra dell'unità.

4. Collegare una cannula nasale o il raccordo di uscita
dell'ossigeno. Assicurarsi che sia ben collegata
prima dell'utilizzo.
NOTA: l'uso di cannule diverse da quelle consigliate può
incidere sulle prestazioni.

b. Far scorrere la batteria verso l'interno fino asentire un clic per confermare l'avvenuto blocco
in sede.

Nota: per il fornitore dell'apparecchiatura: con ilconcentratore di ossigeno FreeStyle Comfort sono
raccomandati per l'uso i seguenti accessori per lasomministrazione dell'ossigeno:• Cannula nasale con tubo da 2,1 m (7 piedi) (L/minmassimi: 6 LPM): Codice CAIRE 5408-SEQ• Ulteriori informazioni sugli accessori consigliati sono
disponibili online sul sito www.caireinc.com.
5. Accendere il sistema FreeStyle Comfort
premendo il pulsante di accensione e selezionare
l'impostazione del flusso prescritta utilizzando
i pulsanti di incremento (+) e decremento (-).

PN MN232-C4 L | Manuale dell’utente IT - 149

FreeStyle Comfort
Frequenza respiratoria
15
20
25
30
35
40

Si udirà un allarme acustico indicante che ildispositivo è acceso.
Nota: all'accensione iniziale del sistema FreeStyle
Comfort, l'impostazione del flusso predefinita è la n. 2.
Se l'impostazione viene modificata, l'unità si accenderà
all'impostazione del flusso precedente.
6. Posizionare la cannula nasale sul viso secondo leistruzioni del produttore fornite con la cannula e
respirare normalmente. Si dovrebbe percepire unsoffio di ossigeno erogato attraverso la cannula
nasale ogni volta che si inala.
7. Per modificare l'impostazione del flusso, premere
il pulsante di incremento (+) o decremento (-) finoa ottenere la portata desiderata. È normale sentire
una differenza di suono quando si modificano leimpostazioni del flusso.

Impostazione volume di impulso non
autoSAT (ml)*
1
2
3
4
5
14,0
10,5
8,4
7,0
6,0
5,3

28,0
21,0
16,8
14,0
12,0
10,5

42,0
31,5
25,2
21,0
18,0
15,8

56,0
42,0
33,6
28,0
24,0
21,0

70,0
52,5
42,0
35,0
30,0
26,3

8. Spegnere il sistema FreeStyle Comfort premendo
il pulsante di accensione/spegnimento.
Nota: lo schermo del display entrerà nella modalità distandby e si spegnerà dopo 20 sec. Premere qualsiasi
pulsante per fare uscire lo schermo dalla modalità di standby.
9. Per le funzionalità avanzate consultare le istruzionidel menu di navigazione

FUNZIONALITÀ SPECIALI DELPRODOTTO
autoSAT®
La funzione autoSAT brevettata da CAIRE permettea FreeStyle Comfort di erogare un volume costante
di ossigeno (bolo) per respiro, fino alla massima
frequenza respiratoria. Man mano che la frequenza
respiratoria dell'utente aumenta, ad esempio durante ladeambulazione, la funzione autoSAT regola il dispositivo
affinché mantenga il volume impostato del bolo.
UltraSense®
Questo sensibile erogatore del dosaggio a impulsi
rileva la variazione di pressione nella respirazione
ed eroga l'impulso di ossigeno durante l'inalazione.

Frequenza respiratoria
15
20
25
30
35
40

Impostazione volume di impulso
autoSAT (ml)*
2
3
4
5

1
10,5
10,5
10,5
10,5
10,5
10,5

21,0
21,0
21,0
21,0
21,0
21,0

31,5
31,5
31,5
31,5
30,0
26,3

42,0
42,0
42,0
35,0
30,0
26,3

*La tolleranza è pari a +/- 15%

150 - IT PN MN232-C4 L | Manuale dell’utente

52,5
52,5
42,0
35,0
30,0
26,3

autoDOSE®
In presenza di un "allarme respiro assente", questa
funzionalità pulsa ossigeno a 20 respiri al minuto,
in base all'impostazione del flusso attuale dell'unità,fino al termine dello stato di allarme.

FreeStyle Comfort
AlimentatoriÈ possibile alimentare il sistema FreeStyle Comfort intre modi:• Gruppo batterie ricaricabile• Alimentazione CA• Alimentazione CC
Le informazioni sul funzionamento di ciascuno di questi
componenti sono riportate nelle sezioni seguenti.
AVVERTENZA: USARE SOLO GLIALIMENTATORI DESCRITTI IN QUESTO
MANUALE DELL'UTENTE. L'UTILIZZO DIALIMENTATORI E GRUPPI BATTERIE NONCAIRE PUÒ CAUSARE DANNI ALL'UNITÀ
E UN FUNZIONAMENTO NON CORRETTO.

RIMOZIONE DELLA BATTERIA

Tirare il pulsante di rilascio della batteria verso ilbasso allontanandolo dal sistema FreeStyle Comfort.
TENENDO PREMUTO questo pulsante, far scorrere
la batteria verso l'esterno estraendola dal concentratore.
AVVERTENZA: IL CONCENTRATOREDI OSSIGENO DEVE ESSERE SPENTOE SCOLLEGATO DALL'ALIMENTAZIONEESTERNA PRIMA DI RIMUOVERE IL
GRUPPO BATTERIE.

GRUPPO BATTERIE

Il sistema FreeStyle Comfort può essere alimentato
dal gruppo batterie ricaricabile che viene fornito con
il dispositivo. Non sono previste limitazioni operativedel concentratore quando è alimentato tramite il
gruppo batterie; sono disponibili tutte le portate con
l'alimentazione a batteria.

CONTROLLO DELLA RICARICA DEL
GRUPPO BATTERIEINSTALLAZIONE DELLA BATTERIA

Allineare la batteria con l'alloggiamento inferiore
dal centro del sistema FreeStyle Comfort. Premere
la batteria verso l'interno fino a sentire un clic per
confermare l'avvenuto blocco in sede.

Per verificare il livello di carica della batteria quando
questa è installata nel sistema FreeStyle Comfort,
fare riferimento alla relativa icona e alla percentualerimanente presenti sul display.
80-100%60-80%

40-60%

20-40%

20%

PN MN232-C4 L | Manuale dell’utente IT - 151

FreeStyle Comfort
Nota: se è collegato l'alimentatore, anziché la percentuale
residua, comparirà un indicatore della spina.
Per controllare la carica della batteria quando questa nonè installata nel sistema FreeStyle Comfort, metterla inpiano e premere il pulsante TEST. Le spie dell'indicatoredella batteria (25-100%) si accendono a destra delpulsante TEST per indicare il livello di carica.

CARICA INIZIALE

La nuova batteria in dotazione con il sistema FreeStyle
Comfort è carica solo parzialmente quando è spedita
dalla fabbrica. Prima di usarla per la prima volta,
è necessario ricaricare completamente il gruppo
batterie utilizzando l'alimentazione CA o CC.

INFORMAZIONI GENERALI DI RICARICA

A meno che non si trovi in modalità aereo, il sistema
FreeStyle Comfort ricarica il gruppo batterie ogni
volta che è installato nel concentratore e quest'ultimo
è collegato a una sorgente di alimentazione CA o CC
esterna.
La batteria è completamente carica quando viene
visualizzato 100%.
In alternativa, è possibile ricaricare il gruppo batterieutilizzando il caricabatterie desktop opzionale.

152 - IT PN MN232-C4 L | Manuale dell’utente

DURATA DEL FUNZIONAMENTOA BATTERIA

Quando l'alimentazione CA o CC esterna è scollegata,il concentratore passerà automaticamente al
funzionamento a batteria se il gruppo batterie
è installato ed è carico. La tabella seguente mostra
le durate tipiche di un gruppo batterie nuovo.
Le durate del gruppo batterie rimarranno costantiindipendentemente dalla frequenza respiratoria.
Impostazione
degli
impulsiDurata batteria
singola (8 celle)

Durata batteria
doppia (16 celle)1fino a 8 ore

fino a 16 ore

2fino a 4 ore

fino a 8 ore

3fino a 3 ore

fino a 6 ore

4fino a 2,25 ore

fino a 4,5 ore

5fino a 2 orefino a 4 ore

TEMPO DI RICARICA TIPICO DELLABATTERIA

Normalmente il gruppo batterie impiega 3,5 ore per
una batteria singola e 6,0 ore per una batteria doppia
per ricaricarsi completamente.
Nota: la batteria inizialmente potrebbe non avviare laricarica se il gruppo batterie è completamente scarico.

DURATA DELLA BATTERIA

Le batterie del sistema FreeStyle Comfort sono ideate
per durare per 300 cicli di carica/scarica.

FreeStyle Comfort
Alimentatore CA e CC
Il sistema Freestyle Comfort è dotato di alimentatori
CA e CC.
L'alimentatore CA include gli elementi seguenti:
	a. Alimentatore CA che funziona tra i 100
e i 240 volt, 50-60 Hz.
	b. Un cavo di alimentazione CA per collegare
l'alimentatore alla presa CA corrispondente.

L'alimentatore CC è composto da un cavo che sicollega direttamente all'unità a un'estremità,e alla presa CC all'altra estremità.

ALIMENTAZIONE CA

2. Collegare il cavo di alimentazione CA
all'alimentatore CA.

L'alimentazione CA è per uso domestico o ovunquesia disponibile una presa di alimentazione standard
CA. Il sistema FreeStyle Comfort è completamentefunzionante con alimentazione CA. Il sistema
FreeStyle Comfort caricherà il gruppo batterie ogni
volta che è disponibile un'alimentazione CA (aeccezione di quando si trova nella modalità aereo).
Il gruppo batterie si ricarica se il sistema FreeStyle
Comfort è acceso o spento.
AVVERTENZA: NON COLLEGAREL'ALIMENTAZIONE ESTERNA SENZA UNGRUPPO BATTERIE INSTALLATO NEL
CONCENTRATORE DI OSSIGENO.
Per collegare il sistema FreeStyle Comfort
all'alimentazione CA:
1. Collegare l'alimentatore CA alla presa delconnettore di alimentazione sulla parte anteriore,in basso a destra del sistema FreeStyle Comfort.
Assicurarsi di spingere verso l'interno in modo chela connessione sia sicura.

3. Collegare l'estremità opposta del cavo dialimentazione CA a una presa a parete o appropriata
di altro tipo.

PN MN232-C4 L | Manuale dell’utente IT - 153

FreeStyle Comfort
4. Accendere il sistema FreeStyle Comfort
premendo il pulsante di accensione e selezionare
l'impostazione del flusso prescritta utilizzando i
pulsanti di incremento (+) e decremento (-).

4. Accendere il sistema FreeStyle Comfort
premendo il pulsante di accensione e selezionare
l'impostazione del flusso prescritta utilizzando
i pulsanti di incremento (+) e decremento (-).
Nota: se il gruppo batterie deve essere ricaricato,l'icona della batteria sul display conterrà una sola barra
arancione.

ALIMENTAZIONE CC

L'alimentazione CC è destinata all'uso nelle prese
accessorie delle automobili, delle barche o di altriveicoli a motore. Il sistema FreeStyle Comfort ècompletamente funzionante con alimentazione CC.
Il sistema FreeStyle Comfort ricarica anche il gruppo
batterie se necessario ogniqualvolta sia disponibile
l'alimentazione CC. La batteria può essere ricaricata
sia se l'apparecchiatura è spenta sia mentre è in uso.
AVVERTENZA: NON COLLEGAREL'ALIMENTAZIONE ESTERNA SENZA UNGRUPPO BATTERIE INSTALLATO NEL
CONCENTRATORE DI OSSIGENO.
1. Accendere il veicolo a motore.
2. Collegare l'alimentatore CC alla presa delconnettore di alimentazione sulla parte anteriore,in basso a destra del sistema FreeStyle Comfort.
Assicurarsi di spingere verso l'interno in modo chela connessione sia sicura.

3. Collegare l'estremità opposta del cavo dialimentazione CC alla presa dell'automobile
o accessoria CC di altro tipo.
154 - IT PN MN232-C4 L | Manuale dell’utente

ATTENZIONE: per evitare possibili incendi
o surriscaldamento, assicurarsi che lapresa di corrente a 12 V CC sia priva
di detriti e che la spina dell'adattatore
dell'alimentatore CC sia inserita
correttamente e rimanga saldamente insede.
ATTENZIONE: le superfici metalliche dell'alimentazione
CC possono diventare MOLTO CALDE quando sono in
uso a causa della corrente elevata. Attendere un tempo
sufficiente affinché le superfici si raffreddino prima di
toccare.
ATTENZIONE: non usare l'alimentatore CC con
prolunghe, splitter o adattatori. L'uso di accessori nonapprovati può causare surriscaldamento o danni.
ATTENZIONE: il collegamento del dispositivo a una
presa di corrente a 12 V CC può scaricare la batteriadel veicolo se il motore non è in funzione, anche se il
dispositivo è spento.
ATTENZIONE: non avviare con i cavi un veicolo mentre
è collegato l'alimentatore CC.

FreeStyle Comfort
Pulizia, cura e manutenzione
ordinaria
MANUTENZIONE ORDINARIA

Manutenzione preventiva
Il dispositivo non richiede manutenzione. I filtri
all'interno del dispositivo sono destinati a durare pertutta la vita utile del dispositivo. Se il dispositivo
non funziona correttamente, l'unità deve essere resa
al fornitore per la riparazione. La manutenzione deicomponenti interni all'interno dell'alloggiamento delsistema FreeStyle Comfort deve essere effettuata dal
fornitore di ossigeno.
L'operatore dell'unità è responsabile della seguente
procedura di manutenzione:• Settimanale – Pulire i filtri di ingresso dell'aria
AVVERTENZA: NON TENTARE DI APRIRE O
RIMUOVERE I COPERCHI DI PLASTICA DELCONCENTRATORE DI OSSIGENO. L'UNITÀ
DEVE ESSERE APERTA SOLO DA UNTECNICO DI ASSISTENZA QUALIFICATO.

FILTRI DI INGRESSO DELL'ARIA

L'aria entra nel sistema FreeStyle Comfort attraverso ifiltri di ingresso dell'aria, in basso nella parte anterioredel concentratore di ossigeno. Questi filtri impediscono
alla polvere e ad altre particelle di grandi dimensioni
presenti nell'aria di entrare nell'unità.
I filtri di ingresso dell'aria vanno puliti almeno una
volta alla settimana. Il fornitore di ossigeno può
consigliare di pulirli più frequentemente, a secondadelle condizioni operative. Attenersi a questa procedura
per pulire correttamente i filtri di ingresso dell'aria.
1. Rimuovere i due filtri dalla parte anteriore delsistema FreeStyle Comfort spingendo verso l'alto la
linguetta corrispondente e tirando verso l'esterno.

AVVERTENZA: NON USARE IL
CONCENTRATORE DI OSSIGENO SENZA
IL FILTRO DI INGRESSO DELL'ARIA
IN POSIZIONE.

CURA DEL GRUPPO BATTERIE
Non esporre il gruppo batterie ai liquidi.

Non tentare di smontare o aprire il gruppo batterie.
Non esporre il gruppo batterie a fiamme libere.
Non far cadere il gruppo batterie.
Tenere il gruppo batterie lontano dai bambini.
Se il gruppo batterie appare danneggiato, rivolgersi al
fornitore di ossigeno prima dell'uso.
Non lasciare mai la batteria all'interno o vicino a un
forno a microonde o ad altri apparecchi di cottura.

CONSERVAZIONE DEL GRUPPOBATTERIE

Non è necessario che la batteria del concentratore
di ossigeno sia completamente scarica prima dellaricarica. Si consiglia di ricaricare la batteria delconcentratore di ossigeno dopo ogni utilizzo.
Non lasciare la batteria in un luogo esposto a calore
eccessivo, come la luce solare diretta o in auto. Ciò
potrebbe incidere negativamente sulle prestazioni
e sulla durata della batteria. Può anche causare
l'emissione di calore, fumo, l'accensione o l'esplosionedella batteria. Il gruppo batterie deve essere conservato
in un ambiente stabile, caratterizzato da bassa umidità(umidità relativa inferiore al 70%) e privo di gas
corrosivi. Per prevenire la ruggine, evitare condizioni
che possono creare condensa come rapide fluttuazionidella temperatura ambiente.
Per la conservazione per periodi pari o superiori a unmese, si raccomanda una carica parziale compresa trail 50% e il 75% (due o tre LED). Una volta ricaricata,
rimuovere la batteria dal concentratore di ossigeno.

2. Lavare i filtri in una soluzione di acqua calda
e sapone.
3. Sciacquare bene i filtri e rimuovere l'acqua ineccesso con un panno morbido privo di pelucchi.
Assicurarsi che i filtri siano completamente asciutti
prima di riposizionarli.
4. Reinserire i filtri asciutti e puliti.
PN MN232-C4 L | Manuale dell’utente IT - 155

FreeStyle Comfort
PULIZIAPrima di svolgere qualsiasi attività di pulizia, spegnere
l'unità e scollegarla dall'alimentazione CA o CC.
NON spruzzare direttamente alcun prodotto sul telaio
esterno. Utilizzare una spugna o un panno umido (nonbagnato). Per pulire l'alloggiamento e gli alimentatori,
spruzzare una soluzione detergente delicata sul pannoo sulla spugna. Procedere come indicato dal produttoredel detergente.
AVVERTENZA: SPEGNERE L'UNITÀ E
SCOLLEGARE IL CAVO DI ALIMENTAZIONEDALLA PRESA ELETTRICA PRIMA DI
PULIRE L'UNITÀ PER PREVENIRE IL
PERICOLO DI SCOSSE ELETTRICHE E
USTIONI ACCIDENTALI. GLI INTERVENTIDI RIPARAZIONE E LA RIMOZIONE DELLE
COPERTURE DEVONO ESSERE ESEGUITIESCLUSIVAMENTE DAL FORNITORE
DELL'APPARECCHIATURA O DA UNTECNICO DI ASSISTENZA QUALIFICATO.
AVVERTENZA: PRESTARE ATTENZIONE A EVITARE
CHE IL CONCENTRATORE DI OSSIGENO E IL
GRUPPO BATTERIE SI BAGNINO O CHE FLUIDIENTRINO NELL'UNITÀ. CIÒ PUÒ PROVOCARE
UN MALFUNZIONAMENTO O UN ARRESTO E UN
AUMENTO DEL RISCHIO DI USTIONI O SCOSSE
ELETTRICHE.
AVVERTENZA: NON USARE LIQUIDI DIRETTAMENTE
SULL'UNITÀ. SEGUE UN ELENCO NON ESAUSTIVO DI
AGENTI CHIMICI SCONSIGLIATI: ALCOL E PRODOTTI
A BASE DI ALCOL, PRODOTTI CONCENTRATI A BASE
DI CLORO (CLORURO DI ETILENE) E PRODOTTI
A BASE OLEOSA. QUESTI NON DEVONO ESSEREUTILIZZATI PER PULIRE L'ALLOGGIAMENTOIN PLASTICA SUL CONCENTRATORE, POICHÉPOSSONO DANNEGGIARE L'UNITÀ.
AVVERTENZA: PULIRE IL CONTENITORE,IL PANNELLO DI CONTROLLO E IL CAVO DI
ALIMENTAZIONE ESCLUSIVAMENTE CON UN
DETERGENTE DELICATO PER USO DOMESTICOAPPLICATO CON UNA SPUGNA O UN PANNOINUMIDITI (NON BAGNATI), QUINDI ASCIUGARETUTTE LE SUPERFICI. EVITARE L'INFILTRAZIONEDI LIQUIDI ALL'INTERNO DEL CONCENTRATORE.
PRESTARE PARTICOLARE ATTENZIONE ALLAPRESA DELL'OSSIGENO PER ASSICURARSI CHEIL COLLEGAMENTO DELLA CANNULA RESTI LIBERO
DA POLVERE, ACQUA E PARTICELLE.

156 - IT PN MN232-C4 L | Manuale dell’utente

AVVERTENZA: UTILIZZARE UNA SPUGNA O UN
PANNO INUMIDITI (NON BAGNATI) PER PULIRE IL
GRUPPO BATTERIE. SPRUZZARE IL DETERGENTE
DELICATO SUL PANNO O SULLA SPUGNA, QUINDI
PULIRE L'ALLOGGIAMENTO DEL GRUPPO BATTERIE.
AVVERTENZA: NON PULIRE CON UN SISTEMA DI
LAVAGGIO AUTOMATIZZATO.

SOSTITUZIONE DELLA CANNULA

Per il corretto utilizzo, attenersi sempre alle istruzionidel produttore della cannula. Sostituire la cannula
nasale o il tubo dell'ossigeno ogni 3-6 mesi (ogni
3 mesi in caso di utilizzo quotidiano) come consigliato
dal produttore della cannula o dal fornitore di ossigeno.
Il medico o il fornitore di ossigeno forniranno leinformazioni riguardanti la pulizia e la sostituzione.
Le forniture aggiuntive per la sostituzione sono
disponibili presso il fornitore di ossigeno.

FreeStyle Comfort
Condizioni di allarmeN. priorità

Descrizione
allarme1

Malfunzionamento
sistema, altatemperaturadella batteria

Display

SegnaleacusticoPiù dettagli / azione suggerita

1 bip ogni
20 secL'unità entra in allarme quando la temperaturadella batteria si sta avvicinando alla soglia
di arresto di sicurezza.
Se viene raggiunta la soglia di arresto,
l'unità si spegne. Se l'unità è collegata
all'alimentazione esterna, appare una
schermata di allarme e l'allarme acusticosi protrae per 2 minuti.
Test con la batteria a temperatura ambiente.
Se il problema persiste, contattare il fornitore.

2

Malfunzionamento
sistema, altatemperatura

1 bip ogni
20 secL'unità entra in allarme quando la temperatura interna si sta avvicinando alla soglia diarresto.
Se viene raggiunta la soglia di arresto, l'unitàsi spegne, appare una schermata di allarme el'allarme acustico si protrae per 2 minuti.
Accertarsi che le ventole dell'unità non siano
ostruite. Se il problema persiste, contattare
il fornitore.

3

Malfunzionamento
sistema, altapressione (altapressione
del sistemao guasto delcompressore)

1 bip ogni
20 sec4

Malfunzionamento sistema, portata
ridotta (valvola
OCD guasta,
ostruzione delpercorso delgas)

1 bip ogni
30 secSe il problema persiste, contattare il fornitore.

5

Arresto perbatteria
scarica

1 bip ogni
30 secL'allarme suona finché la batteria non si
scarica e l'unità si spegne.

Se viene raggiunta la soglia di arresto, l'unitàsi spegne, appare una schermata di allarme el'allarme acustico si protrae per 2 minuti.
Se il problema persiste, contattare il fornitore.

Caricare la batteria.

PN MN232-C4 L | Manuale dell’utente IT - 157

FreeStyle Comfort
N. priorità

Descrizione
allarme6

Nessun respiro
rilevato

Display

Segnaleacustico

Ulteriori dettagli

1 bip ogni
10 minNessun respiro rilevato per 15 sec.
Si attiva dopo il primo respiro o quando
la schermata del riscaldamento è spenta,
a seconda della condizione che si verifichi
per prima.
Controllare che la cannula non presentipieghe o perdite. Aumentare l'impostazione
di sensibilità. Se il problema persiste,contattare il fornitore.

7

Riscaldamento

nessunoL'unità inizia l'auto-somministrazione fino alrilevamento di un respiro o fino alla fine delriscaldamento, a seconda della condizione
che si verifichi per prima.

8

Concentrazione ridotta

1 bip ogni
10 min,per unmassimodi 30 minSe il problema persiste, contattare il fornitore.

9

Avvertenza
batteria
scarica

1 bip,
senza
ripetizione

Caricare la batteria.

10Frequenza
respiratoria
superata

1 bip,
senza
ripetizione

La frequenza respiratoria supera i 40 bpm.

11

Malfunzionamento sistema, guasto
ventolanessunoSe il problema persiste, contattare il fornitore.

12

Malfunzionamento
sistema, bassapressione

nessunoSe il problema persiste, contattare il fornitore.

13

Malfunzionamento
sistema, bassa
temperatura

nessunoLasciare riscaldare l'unità fino a temperatura
ambiente. Se il problema persiste, contattare
il fornitore.

158 - IT PN MN232-C4 L | Manuale dell’utente

Se la somministrazione viene innescata
erroneamente, diminuire l'impostazione
di sensibilità. Se il problema persiste,contattare il fornitore.

FreeStyle Comfort
N. priorità

Descrizione
allarme

Info

Avvio

Display

Segnaleacustico

Ulteriori dettagli

1 bip info

All'accensione appare la schermata e si
dovrebbe sentire un bip che conferma ilfunzionamento del segnalatore acustico.
La schermata di avvio visualizza la versione
software attuale nell'angolo inferiore destro.

Info

Schermata difunzionamento
normale inmodalità autoSAT (nessunaali­mentazione
CC collegata)nessunoInfo

Schermata difunzionamento
normale con
autoSATdisabilitato(nessunaali­mentazione
CC collegata)

NessunoInfo

Impostazione
modificata

1 bip info

Info

Collegare all'alimentazione
esterna nella
condizione
spentanessunoInfo

Collegare all'alimentazione
esterna nella
condizione difunzionamento

nessunoInfo

Scollegare
l'alimentazione
esterna nella
condizione
spenta1 bip info

Appare la schermata CAIRE per 2 secondi,
seguita dallo stato Batteria e alimentazione
per 5 sec prima che lo schermo si spenga.

Appare la schermata CAIRE per 2 sec,
quindi spegnere il display.

PN MN232-C4 L | Manuale dell’utente IT - 159

FreeStyle Comfort
N. priorità

Descrizione
allarme

Display

Info

Scollegare
l'alimentazione
esterna nella
condizione difunzionamento(con batteria
inserita)

1 bip info

Info

Nella modalità
di standby:
quando viene
premuto il
pulsante suo giù, il display
mostra lo statodella batteria
e di carica

nessunoInfo

Modalitàaereo confunzionamento
a batteria

1 bip
lungo

Info

Modalità aereo
con alimentazione di rete

1 bip
lungo

160 - IT PN MN232-C4 L | Manuale dell’utente

Segnaleacustico

Ulteriori dettagli

Appare la schermata per 5 sec,
quindi si spegne.

FreeStyle Comfort
Istruzioni per la navigazione del menu
Si può accedere al Menu in modalità Standby
o Funzionamento. In modalità Funzionamento,
se si verifica un allarme si uscirà automaticamente
dal menu. Se non vengono premuti pulsanti per
20 secondi, si uscirà automaticamente dal menu.
In modalità Standby potrebbe essere necessariopremere il pulsante Menu due volte per riattivare
l'unità, quindi sarà possibile entrare nel Menu.
Sono presenti quattro schermate del menu:
1. Schermata di stato: mostra il codice di rev. delfirmware, il tempo di funzionamento e i cicli discarica della batteria

Modalità aereo

2. Modalità aereo (ON/OFF)
3. Sensibilità: regolabile (1-5)
4. autoSAT (ON/OFF)

Sensibilità

Esempio di schermata di stato

FUNZIONAMENTO DEL MENUIMPOSTAZIONI

autoSATSi accede al menu impostazioni premendo il pulsanteMenu/Seleziona
o Standby.

nelle modalità Funzionamento

Per scorrere fra i menu, premere.

Pagina di stato

PN MN232-C4 L | Manuale dell’utente IT - 161

FreeStyle Comfort
PAGINA DI STATO
La pagina di stato dell'unità Freestyle Comfort mostra
la revisione del firmware, le ore di funzionamento e ilnumero di cicli della batteria sulla batteria collegata.

Per annullare la modalità aereo, premere
è visibile l'icona ANNULLA.

mentreConfermata attivazione della modalità aereo
Pagina di stato

Per le ore di funzionamento vengono mostrate le oreintere.
Le scariche della batteria si riferiscono al numero
di cicli in cui la batteria collegata si è scaricata
e ricaricata.

MODALITÀ AEREO
Il sistema FreeStyle Comfort viene fornito in modalitàaereo disattivata.

Per disattivare la modalità aereo, ripetere i passi
illustrati sopra portando il selettore su OFF.
Premerementre la schermata mostra il simbolo"Avanti" farà avanzare la pagina del menu alla
schermata di opzioni successiva.
La modalità aereo è prevista per l'utilizzo in modalitàdi funzionamento. La modalità aereo verrà disattivata
automaticamente se l'unità viene lasciata in modalitàStandby o con la sola alimentazione a batteria.
Nota: consultare la sezione "Attivare la modalità aereo"
per maggiori informazioni sulla modalità aereo.

MODALITÀ DI SENSIBILITÀ

Schermata principale della modalità aereo

Il sistema FreeStyle Comfort viene fornito con lamodalità di sensibilità preimpostata in fabbrica
sul valore 2, e in condizioni normali non dovrebbe
richiedere alcuna regolazione. L'impostazione disensibilità può essere regolata fra i valori 1-5, dove1 è il valore di sensibilità più basso e 5 il più alto.

Per attivare la modalità, premere
mentre è visibile
l'icona MODIFICA. Il valore si attiverà mostrando laparola "ON" in blu.

Schermata della modalità di sensibilità con impostazione sul valore 2

Selezionata l'attivazione della modalità aereo

Per confermare l'attivazione della modalità aereo,
premerementre è visibile l'icona SALVA. Il valore
si attiverà mostrando "ON" in nero.

162 - IT PN MN232-C4 L | Manuale dell’utenteFreeStyle Comfort
Per regolare la modalità di sensibilità, premere
mentre è visibile l'icona MODIFICA. Il valore passeràal numero in blu successivo.

Selezionata modalità di sensibilità sul valore 3

Per confermare l'impostazione 3 in modalità disensibilità, premere
mentre è visibile l'icona
SALVA. Il valore cambierà mostrando un "3" in nero.

MODALITÀ AUTOSAT
Il sistema FreeStyle Comfort viene fornito conla modalità autoSAT attivata.

Schermata principale della modalità autoSAT

Per disattivare la modalità, premere
mentreè visibile l'icona MODIFICA. Il valore si attiverà
mostrando la parola "OFF" in blu.

Per annullare la modifica della modalità di sensibilità,
premerementre è visibile l'icona ANNULLA.

Selezionata disattivazione della modalità autoSAT

Per confermare la disattivazione della modalità
Confermata la modalità di sensibilità sul valore 3

autoSAT, premere
mentre è visibile l'icona
SALVA. Il valore si attiverà mostrando la parola"OFF" in nero.
Per annullare la modalità autoSAT, premere
mentre è visibile l'icona ANNULLA.

Confermata la modalità di sensibilità sul valore 5 (massimo)

Premerementre la schermata mostra il simbolo"Avanti" farà avanzare la pagina del menu alla
schermata di opzioni successiva.

Confermata disattivazione della modalità autoSAT

Per attivare la modalità autoSAT, ripetere i passi
illustrati sopra portando il selettore su ON.
Premerementre la schermata mostra il simbolo"Avanti" farà avanzare la pagina del menu alla
schermata di opzioni successiva.

PN MN232-C4 L | Manuale dell’utente IT - 163

FreeStyle Comfort
Risoluzione dei problemi
Problema

Probabile causa

Soluzione

FreeStyle Comfort visualizza
un allarme e produce un bip
intermittente.

Fare riferimento alla tabella
Condizioni di allarme.

Fare riferimento alla tabella Condizioni di allarme.

Il concentratore di ossigeno
non si accende.

La batteria è scarica.

Alimentare l'unità mediante una presa CC o CA.

Non è collegato all'alimentazione
esterna.

Assicurarsi che i collegamenti esterni siano saldi.

Malfunzionamento generale.

Eseguire un ciclo di accensione e spegnimento
completo rimuovendo la batteria e scollegando
dalla fonte di alimentazione esterna CA o CC.
Quindi reinserire la batteria; il ciclo di accensione
e spegnimento è completo. Ricollegare alla fonte
di alimentazione esterna, se necessario.
Contattare il fornitore dell'apparecchiatura epassare a un'altra fonte di ossigeno secondo
necessità.

Tutti gli altri problemi.

Eseguire un ciclo di accensione e spegnimento
completo rimuovendo la batteria e scollegando
dalla fonte di alimentazione esterna CA o CC.
Quindi reinserire la batteria; il ciclo di accensione
e spegnimento è completo. Ricollegare alla fonte
di alimentazione esterna, se necessario.
Passare a un'altra fonte di ossigeno disponibile
e contattare il fornitore dell'apparecchiatura.

Nota: alcuni allarmi causano lo spegnimento dell'unità.
Nota: durante un'interruzione di corrente generale, il sistema di allarme si azzera (cancella qualsiasi allarme precedente)e visualizzerà un allarme solo se la condizione sussiste dopo il ripristino dell'alimentazione.

164 - IT PN MN232-C4 L | Manuale dell’utenteFreeStyle Comfort
Informazioni di viaggio

1. Accendere il COP usando SOLO l'alimentazione abatteria, senza collegarlo ad alimentazioni esterne.

Il sistema FreeStyle Comfort è conforme alle linee guida
FAA per l'utilizzo su aeromobili commerciali. Prima delviaggio (aereo, in treno, crociera, ecc.) sarà necessario:• Assicurarsi che il concentratore sia in buone
condizioni di funzionamento.
• Contattare la compagnia aerea almeno 48 ore primadella partenza per informarla che si intende viaggiarecon FreeStyle Comfort.
Si consiglia di portare sempre con sé batterie
aggiuntive durante i viaggi. Per i viaggi in aereo, farein modo di avere sufficiente batteria considerando il
tempo a terra (check-in pre-volo, controlli di sicurezza,ecc.) e le soste. La maggior parte delle compagnie
aeree richiedono di disporre di un numero adeguato di
gruppi batterie completamente carichi per alimentare
il dispositivo per almeno il 150% della durata massima
del viaggio prevista (ore di volo, tempo complessivo
a terra prima e dopo il volo e durante i collegamenti,
nonché per ritardi imprevisti).

2. Abilitare la modalità aereo utilizzando uno dei
seguenti metodi:a. Seguire le istruzioni per attivare la modalità aereo
nella sezione Istruzioni menu di navigazione.

Nota: per i voli si consiglia di trasportare le batterie
di riserva in una borsa per il trasporto sicura.

NORMATIVA FAA

Per evitare cortocircuiti, le batterie di riserva devono
essere protette una a una inserendole nella confezione
originale o isolando in altro modo i terminali, ad esempio
applicando del nastro adesivo sui terminali esposti oinserendo ogni batteria in un sacchetto di plastica o in
una custodia protettiva; inoltre, le batterie devono essere
trasportate esclusivamente nel bagaglio a mano.

b. Premere e mantenere premuti entrambi i pulsantiSu e Giù per 5 secondi finché non si sente un "bip"prolungato e si vede apparire il simbolo dell'aereo.

ATTIVARE LA MODALITÀ AEREO
Alimentazione a batteria
Il COP può essere utilizzato su un aeromobile secondo
le procedure normali SOLO con alimentazione
a batteria, quando non è collegato ad alcuna fonte
di alimentazione esterna.
Alimentazione esterna
Sugli aeromobili, se è disponibile e la sua posizione
è idonea all'uso come fonte di alimentazione per il
COP, è possibile usare una presa elettrica funzionante
come supporto per le batterie del COP. In tal caso,
è necessario seguire questa procedura consigliata per ilpassaggio da alimentazione a batteria ad alimentazione
elettrica dell'aeromobile:

3. Quando appare il simbolo dell'aereo, collegare
il COP.

Nota: a causa delle limitazioni sugli aeromobili, le batterie
non si caricano nella modalità aereo.

PN MN232-C4 L | Manuale dell’utente IT - 165

FreeStyle Comfort
Accessori

INSERIMENTO DELL'UNITÀ IN UNABORSA PER IL TRASPORTO

Per prestazioni e sicurezza corrette, usare
esclusivamente gli accessori in elenco forniti da
CAIRE mediante il fornitore di ossigeno. L'utilizzo diaccessori non presenti nel seguente elenco può influire
negativamente sulle prestazioni e/o sulla sicurezza delconcentratore.

1. Inserire l'unità sul fondo di una borsa, allineando
correttamente le aperture di batteria e filtri.

Accessori standard del sistema FreeStyle Comfort
Borsa per il trasporto

MI396-1

Zaino

MI459-1

Tracolla

MI397-1

Alimentatore CA (tutti i cavi inclusi)

PW036-1S

Cavo di alimentazione CC

CD041-1

Filtro per particelle grosse esterno(destro)FI226-1

Filtro per particelle grosse esterno(sinistro)

FI227-1

2. Chiudere la cerniera nella parte inferiore della borsa.

Accessori opzionali disponibili
Gruppo batterie a 8 celle aggiuntivo

BT034-1S

Gruppo batterie a 16 celle

BT035-1S

Caricabatterie desktop del gruppo
batterie

BT036-1

Borsa per il trasporto degli accessori

MI372-2

Nota: potrebbero essere disponibili opzioni aggiuntive percavi specifici del Paese, dove indicato. Contattare CAIRE
o il fornitore di ossigeno se è necessario ordinare opzioni
alternative.

Istruzioni della borsa peril trasporto
Per inserire e fissare bene l'unità FreeStyle Comfort nellaborsa per il trasporto, usare le seguenti istruzioni.

166 - IT PN MN232-C4 L | Manuale dell’utente

INSTALLAZIONE DELLA TRACOLLA
1. La tracolla può essere fissata sia direttamente
all'unità sia alla borsa per il trasporto.
2. Attaccare la tracolla all'anello sull'unità o sulla borsa.

FreeStyle Comfort
Smaltimento e riutilizzo
Restituire sempre l'unità FreeStyle Comfort, inclusi
tutti i componenti, al fornitore dell'apparecchiatura peril corretto smaltimento. È inoltre possibile rivolgersi
agli uffici locali per istruzioni sul corretto smaltimentodella batteria. L'unità può essere riutilizzata da altre
persone, ma prima deve essere resa al fornitore
dell'apparecchiatura per il ricondizionamento.

RAEE e RoHS
Questo simbolo ha lo scopo di ricordare ai
proprietari dell'apparecchiatura di consegnarla presso
un centro per il riciclaggio al termine della durata utile,
in conformità alla Direttiva sui rifiuti di apparecchiature
elettriche ed elettroniche (RAEE).

Materiali a contatto diretto con
l'operatore• Involucro esterno del concentratore: PC/ABS• Pannello di controllo del concentratore:
poliestere EBG7 o equivalente• Uscita dell'ossigeno, ugello: alluminio• Etichetta dell'unità: Lexan• Connettori dei cavi: policarbonato/cloruro di vinile• Cavo di alimentazione: PVC, metallo• Alimentatore: Lexan 940 (policarbonato)• Gruppo batterie: PC/ABS• Etichette di gruppo batterie e alimentatore:
film di poliestere• Custodia per il trasporto del concentratore:
cordura, neoprene, fodera in nylon• Tracolla: polipropilene, acetile, vinile, cordura,
neoprene

I nostri prodotti sono conformi alle restrizioni dettate
dalla Direttiva sulle sostanze pericolose (RoHS).
È possibile che essi contengano tracce, ma non quantità
rilevanti, di piombo o altri materiali pericolosi.
ATTENZIONE: per il corretto smaltimento,
rivolgersi al fornitore dell'apparecchiaturao all'ente governativo preposto per leistruzioni di smaltimento
ATTENZIONE: smaltire sempre il setaccio molecolare
in conformità a tutte le normative di controllo
ambientale nazionali e locali.

PN MN232-C4 L | Manuale dell’utente IT - 167

FreeStyle Comfort
Test CEM
Linee guida e dichiarazione del produttore – Emissioni elettromagnetiche
Il sistema FreeStyle Comfort è destinato all'uso nell'ambiente elettromagnetico specificato di seguito. Il cliente
o l'utilizzatore del sistema FreeStyle Comfort deve garantirne l'uso all'interno di un ambiente di questo tipo.
Test delle emissioni
Emissioni RF
CISPR 11
Emissioni RF
CISPR 11
Emissioni armoniche
IEC 61000-3-2
Fluttuazioni di tensione/
emissioni di sfarfallii
IEC 61000-3-3

Conformità

Ambiente elettromagnetico - Linee guida

Gruppo 1

Il sistema FreeStyle Comfort impiega energia a radiofrequenza solo
per il funzionamento interno. Pertanto, le emissioni RF sono molto
basse ed è improbabile che possano causare interferenze nelle
apparecchiature elettroniche nelle vicinanze.

Classe B
Classe A
Conforme

168 - IT PN MN232-C4 L | Manuale dell’utente

Il sistema FreeStyle Comfort è idoneo all'uso in tutti gli edifici, compresi
quelli ad uso abitativo e quelli collegati direttamente alla rete pubblica
di alimentazione a bassa tensione utilizzata per rifornire gli edifici ad
uso abitativo.

FreeStyle Comfort
Linee guida e dichiarazione del produttore – Immunità elettromagnetica

Il sistema FreeStyle Comfort è destinato all'uso nell'ambiente elettromagnetico specificato di seguito. Il cliente
o l'utilizzatore del sistema FreeStyle Comfort deve garantirne l'uso all'interno di un ambiente di questo tipo.
Test dell'immunità Livello di test IEC 60601 Livello di conformità
Ambiente elettromagnetico – Linee guida
Ambiente
Il pavimento deve essere di legno, cemento±8 kV per contatto±8 kV per contatto
elettromagnetico –o piastrelle in ceramica. Se i pavimenti
Linee guida
sono ricoperti di materiale sintetico, l'umidità±15 kV per aria
±15 kV per aria
relativa deve essere almeno del 30%.
IEC 61000-4-2
Transitori elettrici
veloci/burst±2 kV per linee
di alimentazione

IEC 61000-4-4±1 kV per linee
di ingresso/uscita±1 kV da linea a linea

Sovratensionetransitoria±2 kV modalità comune
sulle linee CA

IEC 61000-4-5±1 kV differenziale sulle
linee CA

±2 kV per linee dialimentazione
N/D±1 kV da linea a linea

La qualità dell'alimentazione di retedeve corrispondere a quella di una tipica
struttura sanitaria professionale e degli
ambienti sanitari domestici.

La qualità dell'alimentazione di retedeve corrispondere a quella di una tipica
struttura sanitaria professionale e degli
ambienti sanitari domestici.

±2 kV modalità comune
sulle linee I/O esterne
0% UT per 0,5 ciclo
0% UT per 0,5 ciclo(0°, 45°, 90°, 135°, 180°, (0°, 45°, 90°, 135°, 180°,
La qualità dell'alimentazione di rete
225°, 270°, 315°)
225°, 270°, 315°)deve corrispondere a quella di una tipica
Cali di tensione, brevi
struttura sanitaria professionale e degliinterruzioni e variaambienti sanitari domestici. Se l'utilizzatore
0% UT per 1 ciclo (0°)
0% UT per 1 ciclo (0°)
zioni della tensionedel sistema FreeStyle Comfort richiededi alimentazione
continuità di funzionamento in caso disulle linee di ingresso 70% U (30% di calo in
70% UT (30% di calo ininterruzioni dell'alimentazione elettrica,Tdell'alimentazione
UT) per 25/30 cicli (0°)UT) per 25/30 cicli (0°)è consigliabile alimentare il sistema
IEC 61000-4-11
FreeStyle Comfort con un gruppo dicontinuità o una batteria.
0% UT per 250/300 cicli
0% UT per 250/300 cicli
(0°)
(0°)
I campi magnetici alla frequenza diCampo magnetico
alimentazione devono essere ai livelli dialla frequenza di
un luogo tipico in una struttura sanitaria
3 A/m
3 A/m
rete (50/60 Hz)
professionale e degli ambienti sanitari
IEC 61000-4-8
domestici.
NOTA: UT indica la tensione di rete CA prima dell'applicazione del livello di test.
NOTA: in alcuni casi, il sistema Freestyle Comfort può azzerarsi dopo una scarica elettrostatica importante e potrebbe
richiedere un ciclo di accensione e spegnimento completo per riprendere il normale funzionamento. Per ulteriori dettaglivedere la sezione Risoluzione dei problemi.

PN MN232-C4 L | Manuale dell’utente IT - 169

FreeStyle Comfort
Linee guida e dichiarazione del produttore – Immunità elettromagnetica

Il sistema FreeStyle Comfort è destinato all'uso nell'ambiente elettromagnetico specificato di seguito. Il cliente o l'utilizzatoredel sistema FreeStyle Comfort deve garantirne l'uso all'interno di un ambiente di questo tipo.
Test dell'immunità Livello di test IEC 60601 Livello di conformità
Ambiente elettromagnetico – Linee guida
RF condotta
10 Vrms
10 Vrms
Gli apparecchi di comunicazione aradiofrequenza mobili e portatili devono
IEC 61000-4-6
Da 150 kHz a 80 MHz
essere utilizzati a una distanza da qualsiasi
componente del sistema FreeStyle Comfort,
compresi i cavi, superiore alla distanza diseparazione calcolata in base all'equazione
RF irradiata
10 V/m
10 V/mapplicabile alla frequenza del trasmettitore.
IEC 61000-4-3
Da 80 MHz a 2700 MHz
Distanza di separazione consigliatad  1.2 P=

d  1.2 P = Da 80 MHz a 800 MHz=

d  1.2 P = Da 800 MHz a 2700 MHz=

dove P è la potenza di uscita nominalemassima del trasmettitore in watt (W)secondo le indicazioni del produttoredel trasmettitore e d è la distanza diseparazione consigliata in metri (m).
Le intensità dei campi prodotti dai
trasmettitori a radiofrequenza fissi,
determinate mediante un'analisi
elettromagnetica del sito,a devono essere
inferiori al livello di conformità in tutte lebande di frequenza.b
Le interferenze possono verificarsi nei
pressi delle apparecchiature contrassegnateda questo simbolo:

NOTA 1: a 80 MHz e 800 MHz, si applica l'intervallo di frequenza superiore.
NOTA 2: queste linee guida potrebbero non essere applicabili in tutte le situazioni. La propagazione elettromagnetica
è influenzata dall'assorbimento e dalla riflessione di strutture, oggetti e persone.
a
Le intensità del campo prodotte da trasmettitori fissi, quali stazioni base per telefoni via radio (cellulari e cordless) e radio
mobili terrestri, radio amatoriali, trasmissioni radiofoniche AM/FM e trasmissioni televisive, non possono essere previste
con accuratezza a livello teorico. Per valutare l'ambiente elettromagnetico prodotto da trasmettitori RF fissi, è opportuno
eseguire un'indagine elettromagnetica della sede. Se l'intensità di campo misurata nel punto in cui è utilizzato FreeStyle
Comfort supera il livello applicabile di conformità RF sopra indicato, è opportuno appurare che FreeStyle Comfort funzioni
correttamente. Qualora fosse riscontrato un funzionamento fuori dalla norma, potrebbe essere necessario adottare ulteriori
provvedimenti, ad esempio cambiando l'orientamento o la posizione di FreeStyle Comfort.
b

Oltre l'intervallo di frequenza da 150 kHz a 80 MHz, le intensità del campo devono essere inferiori a 10 V/m.

170 - IT PN MN232-C4 L | Manuale dell’utenteFreeStyle Comfort
Distanze di separazione consigliate tra apparecchi per la comunicazione a RF portatilie mobili e FreeStyle Comfort

Il sistema FreeStyle Comfort è destinato all'uso in un ambiente elettromagnetico in cui i disturbi da RF irradiata sono tenuti
sotto controllo. Il cliente o l'utilizzatore del sistema FreeStyle Comfort può prevenire le interferenze elettromagnetichemantenendo una distanza minima tra le apparecchiature di comunicazione RF portatili e mobili (trasmettitori) e ilsistema FreeStyle Comfort come indicato di seguito, calcolando tale distanza in base alla potenza massima di uscita
dell'apparecchiatura di comunicazione.
Potenza di uscita
Distanza di separazione in base alla frequenza del trasmettitore
nominale massimamdel trasmettitore
Da 150 kHz a 80 MHz
Da 80 MHz a 800 MHz
Da 800 MHz a 2,5 GHz
d = 1.2 P
d = 1.2 P
d = 1.2 P
W
0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
Per i trasmettitori con una potenza di uscita massima non elencata sopra, la distanza di separazione consigliata (d) in metri(m) può essere stimata utilizzando l'equazione applicabile alla frequenza del trasmettitore, dove P è la potenza di uscita
massima del trasmettitore in watt (W) ottenuta in base alle informazioni fornite dal produttore.
NOTA 1: a 80 MHz e 800 MHz, si applica la distanza di separazione per l'intervallo di frequenza superiore.
NOTA 2: queste linee guida potrebbero non essere applicabili in tutte le situazioni. La propagazione elettromagnetica
è influenzata dall'assorbimento e dalla riflessione di strutture, oggetti e persone.

Classificazione

Tipo di protezione dalle scosse elettriche:
Classe II	La protezione contro le scosse elettriche è			assicurata dal doppio isolamento.
Grado di protezione dalle scosse elettriche:	Tipo BF Apparecchiatura che fornisce un particolare		grado di protezione dalle scosse elettriche		in relazione a		1) corrente di dispersione ammissibile;		2) affidabilità del collegamento di messa a terradi protezione (se presente).
		Non usare per applicazione cardiaca diretta.
Test indipendente per apparecchiatura elettromedicale
standard:	Testato per la compatibilità alle norme• IEC 60601-1 edizione 3.1: Apparecchi
elettromedicali - Parte 1: Requisiti generali
di sicurezza.
• CAN/CSA C22.2 N. 60601-1-8 edizione 2.1
Apparecchi elettromedicali – Parte 1: Requisiti
generali di sicurezza• FCC Parte 15, Sottoparte B – Classe B
	Radiatori non intenzionali
IP22 - Protezione contro l'ingresso di dita o oggetti
simili di diametro superiore a 12,5 mm. L'acquaa caduta verticale non avrà effetto nocivo se
l'involucro è inclinato con un'angolazione di
15° dalla posizione normale.
Metodo di pulizia e controllo delle infezioni consentiti:
consultare la sezione "Pulizia, cura e manutenzione
corretta" di questo Manuale dell'utente di FreeStyle.
Grado di sicurezza dell'applicazione in presenza di gas
anestetici infiammabili:	apparecchiatura non adatta per questo tipo
di applicazione.
Modalità operativa: funzionamento continuo.

Protezione contro potenziali campi elettromagnetici oaltre interferenze tra l'apparecchiatura e altri dispositivi.
	• Testato per la compatibilità a EN 60601-1-2/IEC 60601-1-2: 4a edizione.
	• Testato per la compatibilità a RTCA/DO160
	Sezione 21 Categoria M.
• CISPR 11 / EN 55011 Classe B Gruppo 1,
	"Apparecchiatura industriale, scientifica e medica(ISM)"PN MN232-C4 L | Manuale dell’utente IT - 171

FreeStyle Comfort
Controlos do utilizador e Indicadores de estado do sistema
ISO 7000
Manual do utilizador; instruções defuncionamento. Reg. n.º 1641

ISO 7010: Símbolos gráficos—Cores e sinalização
de segurança—Sinalização de segurança registada
O manual de instruções deve ser lido.
Reg. n.º M002

O país e a data de fabrico. O "CC"identifica o código do país de duas
letras do país de fabrico. A data defabrico tem o formato AAAA-MM-DD.
Reg. n.º 6049

Manter afastado de chamas abertas,
fogo e faíscas. É proibido fumar e são
proibidas fontes de ignição abertas.
Reg. n.º P003

Manter afastado da chuva, manter seco.
Reg. n.º 0626

Não fume perto da unidade ou quando
a estiver a utilizar. Reg. n.º P002

Limite de empilhamento por número,em que "n" indica o número máximo
de unidades permitido. Reg. n.º 2403
Número de catálogo. Reg. n.º 2493

Aviso. Reg. n.º W001
IEC 60417
Equipamento aplicado tipo BF(grau de proteção contra choques
elétricos). Reg. n.º 5333

Número de série. Reg. n.º 2498
Intervalo de temperatura dearmazenamento ou de funcionamento.
Reg. n.º 0632
Intervalo de humidade dearmazenamento. Reg. n.º 2620
Limites de pressão atmosférica.
Reg. n.º 2621
Nome e morada do fabricante.
Reg. n.º 3082

Equipamento de Classe II, com
Isolamento duplo Reg. n.º 5172
Regulamento (UE) 2017/745
Representante autorizado naUnião Europeia
Se o rótulo de identificação única
do dispositivo (UDI) do produto tivero símbolo CE####, o dispositivo está
em conformidade com os requisitos
relativos aos dispositivos médicos.
O símbolo CE#### indica o número
do organismo notificado.
Símbolos adicionais

Este lado para cima. Reg. n.º 0623

Manter afastado de materiais
inflamáveis, óleo e gorduras.

Frágil, manusear com cuidado.
Reg. n.º 0621

Não desmontar.

Contém substâncias perigosas.
Reg. n.º 3723

Aumentar o fluxo.

Importador. Reg. n.º 3725

Reduzir o fluxo.

Saída de O2Botão de menu.
Interruptor de alimentação.
Reg. n.º 5009
Em conformidade com
a RTCA/DO160 Secção
21 Categoria M

172 - PT PN MN232-C4 L | Manual do usuário

O fabricante deste POC determinou que
este dispositivo está em conformidade
com todos os critérios de aceitação
da FAA aplicáveis e com a utilização
a bordo de aeronaves.

FreeStyle Comfort
O fabricante deste POC determinou que
este dispositivo está em conformidade
com todos os critérios de aceitação
da FAA aplicáveis e com a utilização
a bordo de aeronaves.

Diretiva do Conselho 2012/19/UE
REEE
Este símbolo destina-se a recordar os
proprietários do equipamento de que
devem entregá-lo a uma instalação dereciclagem no final da sua vida útil,de acordo com a Diretiva relativa aos
Resíduos de Equipamentos Elétricose Eletrónicos (REEE). Os nossos
produtos estão em conformidade com
a diretiva de Restrição de Substâncias
Perigosas (RoHS). Não contêm mais
do que quantidades residuais de chumbo
ou de outros materiais perigosos.

Símbolo de Reciclagem
Representante autorizado na Suíça(ADD)
Se o dispositivo possuir a marca UKCA,
tal como indicado, com UKCA#### a
indicar o número do organismo notificado,
este dispositivo está em conformidade
com os regulamentos UKCA.
21 CFR 801.15
A lei federal restringe a venda deste
dispositivo a ou por ordem de um médico.

IEC 60601-1-8
Alarme técnico de baixa prioridade

Certificação ETL
Certificado para os mercados dos EUA
e do Canadá, em conformidade com asnormas aplicáveis dos EUA e do Canadá
Contém a ID da FCC: WAP2001
Aviso da FCC
Este dispositivo pode conter CYBLE-022001-00,
incluindo a antena 2450AT18B100 da Johnson
Technology, em conformidade com a parte 15 dos
regulamentos da FCC. O dispositivo cumpre os
requisitos de aprovação para transmissor modular
conforme descrito na nota pública da FCC,
DA00-1407. O funcionamento do transmissor
está sujeito às duas condições seguintes (1)O dispositivo não pode causar interferências
nocivas e (2) Este dispositivo deve aceitar quaisquerinterferências recebidas, incluindo interferências
que possam causar um funcionamento indesejado.
O produto pode estar abrangido por uma ou
mais patentes, dos EUA e internacionais.
Visite o nosso website abaixo para obter uma
lista das patentes aplicáveis. Patente:www.caireinc.com/corporate/patents
Símbolos gráficos para utilização no equipamento
IEC 60417

ISO 15223-1
Dispositivo médico. Reg. n.º 5.7.7
Identificação única do dispositivo
n.º 5.7.10
A S/NZS 4417.1, AS/NZS 4417.2
Marca de conformidade regulamentar
da ACMA – indica a conformidade com
os regulamentos da ACMA (ADD)
Tecnologia sem fios
Tipos de ligação

SSP, iAP2, GATT

Frequência

2402 a 2480 MHzPotência máxima de saídade RF+4 dBmIntervalo de funcionamento

10 m (Classe 2)

Recomenda-se que o aparelho seja colocado a uma
distância mínima de 1,1 cm (0,43 pol.) do corpo duranteo funcionamento. Não aplicável a máscaras, tubos dear ou acessórios.

Corrente contínua Reg. n.º 5031
Corrente alternada Reg. n.º 5032
IEC 60601-1
À prova de gotas

PN MN232-C4 L | Manual do usuário PT - 173

FreeStyle Comfort
Desembalar o FreeStyle Comfort
Verifique se todos oscomponentes listados estão
incluídos na embalagem. Se
faltar algum artigo, contacte o
seu fornecedor de oxigénio deimediato.

Acessórios padrão
• Concentrador de oxigénio portátil• Bolsa de transporte• Alça a tiracolo• Unidade da bateria de 8 células• Fonte de alimentação CA• Cabo de alimentação CA• Transformador de CC• Manual do utilizador

Acessórios opcionais• Unidade da bateria de 8 células
adicional• Unidade da bateria de 16 células• Carregador de bancada• Bolsa de transporte para todos
os acessórios

Conhecer o FreeStyle Comfort
Familiarize-se com as principais características do FreeStyle Comfort e da sua interface do utilizador.

Vista frontal
Fixação da alçaInterface doutilizador

Saída deOxigénio
Saída de ar

Admissão
de ar/filtros
Admissão de ar/filtros
Patilha dorecetáculo
da porta dealimentação
Conjunto
de bateriaBotão de libertação
da unidade da bateria
174 - PT PN MN232-C4 L | Manual do usuário

FreeStyle Comfort
Vista superiorEcrã do visor
Aumentaro fluxoBotão de energiaBotão de menu

Reduzir o fluxoEcrã do visor alimentado por bateria
Definição
de fluxo

Visualização
da bateria

Indicador autoSAT
Percentagem
da bateria
restante

Visualização
do alarme
Indicador deadministração
do boloEcrã do visor na alimentação CA/CC

Definição
de fluxo

Espaço do visorde alarmes

Visualização
da bateria

Indicador de carga

PN MN232-C4 L | Manual do usuário PT - 175

FreeStyle Comfort
Descrição do Concentradorde Oxigénio
O ar que respiramos contém cerca de 21% de oxigénio,78% de azoto, e 1% de outros gases. No FreeStyle®
Comfort®, o ar ambiente é aspirado pela máquina
através das entradas de ar. Em seguida, passa atravésde um material absorvente chamado peneira molecular.
Este material separa o oxigénio do azoto e permite
apenas a passagem do oxigénio. O resultado é um fluxode oxigénio de elevada concentração administrado ao
utilizador.
O FreeStyle Comfort combina tecnologias avançadas
de concentração e conservação de oxigénio para
criar um dispositivo portátil que permite a máxima
portabilidade e mobilidade. A administração por fluxo
pulsátil avançada deteta rapidamente a inspiração doutilizador e administra oxigénio no início de cadainalação. A administração por fluxo pulsátil é ideal para
atividades e períodos passados no exterior. Este tipo deadministração conserva oxigénio e prolonga a vida útilda bateria, ao contrário do fluxo contínuo. O FreeStyle
Comfort é suficientemente leve para ser transportado
com uma alça de tiracolo. Pode ser alimentado por CA(parede), CC (veículo), ou bateria.

Uso Pretendido
O Concentrador de oxigénio CAIRE FreeStyle
Comfort destina-se à administração de oxigénio
suplementar. O dispositivo não foi concebido como
aparelho de suporte de vida, nem possui quaisquer
funcionalidades de monitorização do doente.

Por que motivo o seu médico lhe
prescreveu oxigénio
Muitas pessoas sofrem de uma variedade de doenças
cardíacas, pulmonares e outras patologias respiratórias.
Um número significativo dessas pessoas pode
beneficiar de terapias de oxigénio suplementar em
casa, quando viajam, ou enquanto participam em
atividade diárias fora de casa.
O oxigénio é um gás que compõem 21% do arambiente que respiramos e os nossos corpos dependem
de um fornecimento estável para que funcionem
adequadamente. O seu médico prescreveu um fluxo
ou definição para atender à sua condição respiratória
específica.

176 - PT PN MN232-C4 L | Manual do usuário

Embora o oxigénio não cause dependência, as terapias
com oxigénio não autorizadas podem ser perigosas.
Deve procurar aconselhamento médico antes de usar este
concentrador de oxigénio. O Fornecedor de equipamentos
que lhe forneceu o equipamento de oxigénio demonstrarácomo definir a taxa de fluxo prescrita.

Perfil do operador
Os concentradores destinam-se a fornecer oxigénio
suplementar a utilizadores que sofrem de desconforto
devido a patologias que afetam a capacidade dospulmões de transferir o oxigénio no ar para a corrente
sanguínea. Os concentradores de oxigénio portáteis
(POCs) não armazenam nem contêm oxigénio.
Não necessitam de reabastecimento e podem serrecarregados em qualquer lado onde exista alimentação
CA ou CC disponível (exceto numa aeronave).
Isto torna o utilizador relativamente autossuficiente
em termos de utilização doméstica, em ambulatório(no interior e exterior da casa) e facilita a mobilidade
e estilo de vida geral. A utilização do concentradorde oxigénio requer a prescrição de um médico.
O utilizador típico da terapia de oxigénio tem uma
idade superior a 65 anos de idade e sofre de uma
variedade de doenças respiratórias, incluindo a Doença
Pulmonar Obstrutiva Crónica (DPOC). Geralmente,
os utilizadores têm boas capacidades cognitivas e
têm de ser capazes de comunicar o seu desconforto.
Se o utilizador não for capaz de comunicar o seu
desconforto, ou não for capaz de ler e compreenderos rótulos e instruções de utilização do concentrador,a utilização do mesmo só é recomendada sob a
supervisão de alguém com capacidade para tal.
Em caso de desconforto ao utilizar o concentrador,o utilizador deve contactar o seu prestador de cuidadosde saúde. Recomenda-se também aos utilizadores
que disponham de uma fonte de oxigénio de reserva(ou seja, um cilindro de oxigénio) para a eventualidadede uma falha de energia ou falha do concentrador.
A utilização do concentrador não requer outras
aptidões ou capacidades especiais.

FreeStyle Comfort
Importante!
As instruções de segurança estão definidas da seguinte
forma:AVISO: INFORMAÇÕES DE SEGURANÇA
IMPORTANTES RELATIVAS A PERIGOSQUE PODEM CAUSAR LESÕES GRAVES.
ATENÇÃO: informações importantes paraevitar danos no FreeStyle Comfort.
Nota: informações que necessitam de atenção especial.

Indicações de UtilizaçãoAVISO: É MUITO IMPORTANTESELECIONAR APENAS O NÍVEL DEOXIGÉNIO PRESCRITO. NÃO ALTERE
A SELEÇÃO DO FLUXO, EXCETO SE TIVER
RECEBIDO INDICAÇÕES NESSE SENTIDOPOR PARTE DE UM MÉDICO QUALIFICADO.
AVISO: ESTE DISPOSITIVO NÃO FOI TESTADOOU APROVADO PARA UTILIZAÇÃO EM DOENTESPEDIÁTRICOS.
AVISO: A LEI FEDERAL (EUA) RESTRINGE A VENDAOU ALUGUER DESTE DISPOSITIVO À CLASSEMÉDICA OU A OUTROS PROFISSIONAIS DE SAÚDEQUALIFICADOS.
AVISO: ESTA UNIDADE NÃO DEVE SER USADA COMOSUPORTE DE VIDA. PODE SER NECESSÁRIA UMA
MONITORIZAÇÃO ADICIONAL PARA QUALQUERUTILIZADOR QUE NÃO POSSA COMUNICARDESCONFORTO DURANTE A UTILIZAÇÃO DESTEDISPOSITIVO. OS UTILIZADORES COM INSUFICIÊNCIA
AUDITIVA E/OU VISUAL PODEM NECESSITAR DEAJUDA COM OS ALARMES DE MONITORIZAÇÃO.
AVISO: AS CONFIGURAÇÕES DO MODO DE
DOSAGEM PULSÁTIL DEVEM SER DETERMINADAS
INDIVIDUALMENTE PARA CADA UTILIZADOR,DE ACORDO COM AS SUAS NECESSIDADES EMREPOUSO, DURANTE O EXERCÍCIO E DURANTEVIAGENS.
AVISO: ESTE DISPOSITIVO DESTINA-SE A SERUTILIZADO EM CUIDADOS DE SAÚDE AO DOMICÍLIO
E EM AMBIENTES AMBULATÓRIOS SEMELHANTES.

AVISO: ALGUNS ESFORÇOS RESPIRATÓRIOS DO
DOENTE PODEM NÃO ACIONAR A FUNÇÃO DE
CONSERVAÇÃO.
AVISO: O FREESTYLE COMFORT NÃO SE DESTINAA UTILIZAÇÃO NUM DOENTE TRAQUEOSTOMIZADO.
AVISO: AS MULHERES GRÁVIDAS OU A AMAMENTAR
NÃO DEVEM UTILIZAR OS ACESSÓRIOS
RECOMENDADOS NESTE MANUAL, UMA VEZ QUECONTÊM FTALATOS.

Contraindicações de utilizaçãoAVISO: EM DETERMINADAS
CIRCUNSTÂNCIAS, A UTILIZAÇÃO DEOXIGÉNIO SEM PRESCRIÇÃO MÉDICA
PODE SER PERIGOSA. ESTE APARELHODEVE SER UTILIZADO APENAS QUANDOPRESCRITO POR UM MÉDICO.
AVISO: ESTE DISPOSITIVO NÃO SE DESTINA ASER UTILIZADO EM AMBIENTES DE RESSONÂNCIAMAGNÉTICA OU NA SUA PROXIMIDADE, TAIS COMO
MÁQUINAS DE RESSONÂNCIA MAGNÉTICA.
AVISO: NÃO DEVE SER UTILIZADO NA PRESENÇA
DE ANESTÉSICOS INFLAMÁVEIS.
AVISO: TAL COMO ACONTECE COM QUALQUEROUTRO APARELHO ALIMENTADO A ENERGIAELÉTRICA, O UTILIZADOR PODE EXPERIENCIAR
PERÍODOS DE NÃO FUNCIONAMENTO COMORESULTADO DE INTERRUPÇÕES DA ENERGIAELÉTRICA, OU NECESSITAR QUE O CONCENTRADORDE OXIGÉNIO SEJA REPARADO POR UM TÉCNICOQUALIFICADO. O CONCENTRADOR DE OXIGÉNIONÃO É ADEQUADO PARA UTILIZADORES QUEPOSSAM SOFRER CONSEQUÊNCIAS ADVERSAS
PARA A SAÚDE COMO RESULTADO DE TAIS
INTERRUPÇÕES TEMPORÁRIAS.
AVISO: ESTE DISPOSITIVO NÃO ESTÁ APROVADOPARA UTILIZAÇÃO EM TODOS OS AMBIENTES.
A EXPOSIÇÃO A DETERMINADAS FONTESELETROMAGNÉTICAS PODE RESULTAR EM
INTERFERÊNCIAS OU NUM DESEMPENHO NÃO
INTENCIONAL DO DISPOSITIVO. EM CASO DE
INTERFERÊNCIA, O APARELHO DEVE SERDESLIGADO IMEDIATAMENTE.

AVISO: ESTE DISPOSITIVO DESTINA-SE A SERUTILIZADO PELO DOENTE QUE ESTÁ A RECEBEROXIGENOTERAPIA.

PN MN232-C4 L | Manual do usuário PT - 177

FreeStyle Comfort
Diretrizes de Segurança
AVISO: É PROIBIDA QUALQUER
MODIFICAÇÃO A ESTE EQUIPAMENTO.
AVISO: O DISPOSITIVO FORNECE OXIGÉNIO DE
ELEVADA CONCENTRAÇÃO QUE PROMOVE UMACOMBUSTÃO RÁPIDA. NÃO AUTORIZE QUE SE
FUME OU QUE SE FAÇAM CHAMAS DENTRO DAMESMA DIVISÃO (1) DESTE DISPOSITIVO OU (2) DEQUALQUER ACESSÓRIO PORTADOR DE OXIGÉNIO.
O NÃO CUMPRIMENTO DESTE AVISO PODEORIGINAR UM INCÊNDIO GRAVE, DANOS MATERIAISE/OU PROVOCAR LESÕES FÍSICAS OU MORTE.
AVISO: NÃO UTILIZE A UNIDADE NUM ESPAÇO
RESTRITO OU CONFINADO (OU SEJA, NUMACAIXA PEQUENA OU NUMA BOLSA) ONDE A
VENTILAÇÃO POSSA FICAR LIMITADA. ESTA AÇÃOPODE PROVOCAR O SOBREAQUECIMENTO DO
CONCENTRADOR DE OXIGÉNIO E PREJUDICARO SEU DESEMPENHO.
AVISO: O CONCENTRADOR DEVE SER COLOCADONUM LOCAL ONDE SEJA EVITADA A ENTRADA DE
FUMO, POLUENTES OU VAPORES.
AVISO: NO CASO DE OCORRER UM INCIDENTEGRAVE COM ESTE DISPOSITIVO, O UTILIZADORDEVE COMUNICAR IMEDIATAMENTE O INCIDENTE AOFORNECEDOR E/OU AO FABRICANTE. UM INCIDENTEGRAVE É DEFINIDO COMO UMA LESÃO, MORTE,OU O POTENCIAL PARA CAUSAR LESÕES/MORTE,CASO OCORRA UMA REINCIDÊNCIA DO INCIDENTE.
O UTILIZADOR PODE TAMBÉM COMUNICAR OINCIDENTE À AUTORIDADE COMPETENTE DO PAÍSONDE O INCIDENTE OCORREU.
AVISO: A UTILIZAÇÃO DE ALGUNSACESSÓRIOS DE ADMINISTRAÇÃODE OXIGÉNIO, NÃO ESPECIFICADOSPARA ESTE CONCENTRADOR, PODEPREJUDICAR O SEU DESEMPENHO.
OS ACESSÓRIOS RECOMENDADOSESTÃO MENCIONADOS NESTE MANUAL.
AVISO: SE O CONCENTRADOR DE OXIGÉNIO TIVERCAÍDO, ESTIVER DANIFICADO OU TIVER SIDOEXPOSTO A ÁGUA, ENTRE EM CONTACTO COM OFORNECEDOR DO EQUIPAMENTO PARA INSPEÇÃOOU POSSÍVEL REPARAÇÃO DO DISPOSITIVO. NÃO
USE O CONCENTRADOR DE OXIGÉNIO CASO ESTEAPRESENTE O CABO DE ALIMENTAÇÃO OU FICHADANIFICADOS.
AVISO: PRENDA DEVIDAMENTE O CONCENTRADORDE OXIGÉNIO QUANDO O TRANSPORTAR NUM
VEÍCULO, COM O CINTO OU DE OUTRO MODO,PARA PREVENIR DANOS OU LESÕES.

178 - PT PN MN232-C4 L | Manual do usuário

AVISO: OS REGULAMENTOS DO DEPARTAMENTONORTE-AMERICANO DOS TRANSPORTES (DOT) E
DAS NAÇÕES UNIDAS (NU) EXIGEM A REMOÇÃODA BATERIA DO APARELHO EM TODAS AS
VIAGENS AÉREAS INTERNACIONAIS QUANDO OCONCENTRADOR DE OXIGÉNIO É REGISTADO
COMO BAGAGEM. PARA TRANSPORTAR OCONCENTRADOR DE OXIGÉNIO, A BATERIATAMBÉM TEM DE SER REMOVIDA DO DISPOSITIVO E
EMBALADA ADEQUADAMENTE.
AVISO: NÃO UTILIZE ESTE EQUIPAMENTO SEM
PRIMEIRO LER E COMPREENDER ESTE MANUAL.
CASO NÃO PERCEBA OS AVISOS E AS INSTRUÇÕES,CONTACTE O SEU FORNECEDOR DO EQUIPAMENTOANTES DE TENTAR USÁ-LO; CASO CONTRÁRIO,PODEM OCORRER LESÕES OU DANOS.
AVISO: SE SENTIR ALGUM DESCONFORTO OU SEENCONTRAR NUMA SITUAÇÃO DE EMERGÊNCIA
MÉDICA, PROCURE ASSISTÊNCIA MÉDICA DE
IMEDIATO.
AVISO: UTILIZAR O CONCENTRADOR DE OXIGÉNIOFORA DOS LIMITES DE TEMPERATURA DEFUNCIONAMENTO PODE LIMITAR A CAPACIDADE
DO CONCENTRADOR DE CUMPRIR COM A
ESPECIFICAÇÃO DA CONCENTRAÇÃO DE OXIGÉNIO.
CONSULTE A SECÇÃO RELATIVA A ESPECIFICAÇÕES
DESTE MANUAL PARA CONSULTAR OS LIMITES DETEMPERATURA.
AVISO: NÃO UTILIZE LUBRIFICANTES, GORDURAS,
PRODUTOS DERIVADOS DO PETRÓLEO OUINFLAMÁVEIS COM OS ACESSÓRIOS PORTADORESDE OXIGÉNIO, OU COM O CONCENTRADOR DEOXIGÉNIO. DEVEM SER USADAS APENAS LOÇÕESOU UNGUENTOS À BASE DE ÁGUA, COMPATÍVEISCOM OXIGÉNIO. O OXIGÉNIO ACELERA ACOMBUSTÃO DE SUBSTÂNCIAS INFLAMÁVEIS.
AVISO: A UTILIZAÇÃO INCORRETA DA BATERIA PODEPROVOCAR O AQUECIMENTO E INFLAMAÇÃO DA
MESMA E CAUSAR LESÕES GRAVES. CERTIFIQUESE DE NÃO PERFURAR, ATINGIR, PISAR OU DEIXAR
CAIR A BATERIA, OU DE A SUBMETER A IMPACTOSOU CHOQUES FORTES. A UTILIZAÇÃO DE UMA
BATERIA DANIFICADA PODE PROVOCAR LESÕES.
AVISO: NÃO PROVOQUE UM CURTO-CIRCUITO NOSCONTACTOS DE METAL DA BATERIA COM OBJETOSMETÁLICOS, COMO CHAVES OU MOEDAS. ISTO
PODE RESULTAR EM FAÍSCAS OU NA GERAÇÃO DE
UM CALOR EXCESSIVO.

FreeStyle Comfort
AVISO: DURANTE A UTILIZAÇÃO DO
CONCENTRADOR DE OXIGÉNIO PORTÁTIL NO
EXTERIOR COM A FONTE DE ALIMENTAÇÃO
CA, LIGUE A FONTE DE ALIMENTAÇÃO
EXCLUSIVAMENTE A UMA TOMADA PROTEGIDACOM UM DISJUNTOR DIFERENCIAL RESIDUAL (DR).
AVISO: O CONCENTRADOR DE OXIGÉNIO NÃO DEVESER ENCOSTADO OU EMPILHADO COM OUTROSDISPOSITIVOS. SE NÃO FOR POSSÍVEL EVITARENCOSTAR OU EMPILHAR O DISPOSITIVO, ESTEDEVE SER SUPERVISIONADO A FIM DE GARANTIRO SEU NORMAL FUNCIONAMENTO.
AVISO: A UTILIZAÇÃO DE CABOS E ADAPTADORESQUE NÃO OS ESPECIFICADOS, À EXCEÇÃODOS CABOS E ADAPTADORES VENDIDOS PELOFABRICANTE DO EQUIPAMENTO ENQUANTO
PEÇAS DE SUBSTITUIÇÃO PARA COMPONENTESINTERNOS, PODE RESULTAR NO AUMENTO DASEMISSÕES OU NUMA DIMINUIÇÃO DA IMUNIDADE
DO CONCENTRADOR DE OXIGÉNIO.
AVISO: UTILIZE APENAS A TENSÃO ELÉTRICAESPECIFICADA NA ETIQUETA DE ESPECIFICAÇÕESAFIXADA NO DISPOSITIVO.
AVISO: NÃO UTILIZE EXTENSÕES COM ESTAUNIDADE OU LIGUE VÁRIAS FICHAS À MESMATOMADA ELÉTRICA. A UTILIZAÇÃO DE EXTENSÕES
PODE AFETAR NEGATIVAMENTE O DESEMPENHODO DISPOSITIVO. A EXISTÊNCIA DE DEMASIADASFICHAS NUMA TOMADA PODE RESULTAR NUMASOBRECARGA DO QUADRO ELÉTRICO, LEVANDOÀ ATIVAÇÃO DO DISJUNTOR/FUSÍVEL OU A UM
INCÊNDIO CASO O DISJUNTOR OU O FUSÍVEL
DEIXEM DE FUNCIONAR.
ATENÇÃO: na eventualidade de um alarmeou se observar que o concentrador deoxigénio não está a funcionar corretamente,
consulte a secção de Resolução deproblemas neste manual. Se não conseguir
resolver o problema, contacte o seu
Fornecedor do Equipamento.

AVISO: NÃO PROCEDA A QUALQUER
TIPO DE MANUTENÇÃO PARA ALÉM DASSOLUÇÕES POSSÍVEIS APRESENTADASNESTE MANUAL.
AVISO: UTILIZE APENAS ACESSÓRIOS
RECOMENDADOS PELO FABRICANTE. A UTILIZAÇÃODE OUTROS ACESSÓRIOS PODE SER PERIGOSA,CAUSAR GRAVES DANOS NO SEU CONCENTRADORDE OXIGÉNIO E INVALIDAR A GARANTIA.
AVISO: O FABRICANTE RECOMENDA UMA FONTE
ALTERNATIVA DE OXIGÉNIO SUPLEMENTAR NA
EVENTUALIDADE DE UMA FALHA DE ENERGIA,CONDIÇÃO DE ALARME OU AVARIA MECÂNICA.
CONSULTE O SEU MÉDICO OU FORNECEDOR DO
EQUIPAMENTO PARA INFORMAÇÕES SOBRE O TIPO
DE SISTEMA DE RESERVA REQUERIDO.
AVISO: NÃO PERMITA QUE A ENTRADA E A SAÍDA
DE AR FIQUEM BLOQUEADAS. NÃO DEIXE CAIR NEM
INSIRA QUAISQUER OBJETOS NAS ABERTURAS
DO DISPOSITIVO. ESTA AÇÃO PODE PROVOCARO SOBREAQUECIMENTO DO CONCENTRADOR DEOXIGÉNIO E PREJUDICAR O SEU DESEMPENHO.
AVISO: FUMAR DURANTE A UTILIZAÇÃO DEOXIGÉNIO É A PRIMEIRA CAUSA DE LESÕES
PROVOCADAS POR FOGO E MORTESRELACIONADAS. DEVERÁ CUMPRIR OS SEGUINTESAVISOS DE SEGURANÇA:AVISO: NÃO AUTORIZE QUE SE FUME, QUE SE
ACENDAM VELAS, OU SE FAÇAM CHAMAS NA MESMADIVISÃO ONDE SE ENCONTRA O DISPOSITIVO OU OSACESSÓRIOS PORTADORES DE OXIGÉNIO.
AVISO: FUMAR DURANTE A UTILIZAÇÃO DE
UMA CÂNULA DE OXIGÉNIO PODE PROVOCARQUEIMADURAS FACIAIS E POSSIVELMENTE A MORTE.
AVISO: CASO SEJA FUMADOR, OS 3 PASSOSSEGUINTES PODEM SALVAR-LHE A VIDA: DESLIGUEO CONCENTRADOR DE OXIGÉNIO, RETIRE ACÂNULA E SAIA DA DIVISÃO NA QUAL SE ENCONTRAO DISPOSITIVO.

ATENÇÃO: para evitar a anulação da garantia,
siga todas as instruções dos fabricantes.

PN MN232-C4 L | Manual do usuário PT - 179

FreeStyle Comfort
AVISO: OS DÍSTICOS “PROIBIDO FUMAR – OXIGÉNIOEM UTILIZAÇÃO” DEVEM ESTAR BEM VISÍVEIS
NA CASA OU NO LOCAL EM QUE O OXIGÉNIOESTÁ A SER UTILIZADO. OS UTILIZADORES E OS
SEUS PRESTADORES DE CUIDADOS TÊM DE SERINFORMADOS ACERCA DOS PERIGOS DE FUMAR
NA PRESENÇA DE OU DURANTE A UTILIZAÇÃO DEOXIGÉNIO MEDICINAL.
AVISO: NÃO UTILIZE O CONCENTRADOR DEOXIGÉNIO NA PRESENÇA DE GASES INFLAMÁVEIS,SOB RISCO DE UMA COMBUSTÃO RÁPIDA
PODER PROVOCAR DANOS MATERIAIS, LESÕESCORPORAIS OU MORTE.
AVISO: REMOVER A CÂNULA E COLOCÁ-LA NO
VESTUÁRIO, ROUPA DE CAMA, SOFÁS OU OUTRO
MATERIAL ALMOFADADO IRÁ DEFLAGRAR UMINCÊNDIO QUANDO EXPOSTOS A UM CIGARRO,FONTE DE CALOR OU CHAMA.
AVISO: NUNCA DEIXE A CÂNULA NASAL EM CIMA/
BAIXO DE PEÇAS DE ROUPA, COLCHAS OU
ALMOFADAS DE CADEIRAS. SE A UNIDADE ESTIVER
LIGADA MAS NÃO ESTIVER EM UTILIZAÇÃO, O
OXIGÉNIO TORNARÁ O MATERIAL INFLAMÁVEL.
CERTIFIQUE-SE DE QUE A ALIMENTAÇÃO ESTÁDESLIGADA (OFF) QUANDO O CONCENTRADOR DEOXIGÉNIO NÃO ESTIVER A SER UTILIZADO.
AVISO: OS ACESSÓRIOS DE FORNECIMENTO DEOXIGÉNIO APROVADOS SÃO CONCEBIDOS PARAREDUZIR A PROPAGAÇÃO DE INCÊNDIOS PARASEGURANÇA DO DOENTE. A UTILIZAÇÃO DE
ACESSÓRIOS NÃO APROVADOS PODE AUMENTARO RISCO DE LESÕES OU DANOS RELACIONADOS
COM INCÊNDIOS.
AVISO: O CONCENTRADOR DE OXIGÉNIO DEVEESTAR DESLIGADO E DESCONECTADO DA
ALIMENTAÇÃO EXTERNA ANTES DE SER REMOVIDAA UNIDADE DA BATERIA.
AVISO: AS PEÇAS PEQUENAS PODEM
REPRESENTAR UM POTENCIAL PERIGODE ASFIXIA.
AVISO: EVITE O CONTACTO COM A ABERTURA DESAÍDA DO AR DURANTE MAIS DE 10 SEGUNDOS UMA
VEZ QUE PODE ESTAR QUENTE.
AVISO: O VENTO OU CORRENTES DE AR FORTESPODEM AFETAR ADVERSAMENTE A ADMINISTRAÇÃOPRECISA DA TERAPIA DE OXIGÉNIO.
AVISO: ESTE PRODUTO PODE EXPÔ-LO A
SUBSTÂNCIAS QUÍMICAS COMO O NÍQUEL, QUE ÉCONSIDERADO UM AGENTE CANCERÍGENO PELOESTADO DA CALIFÓRNIA. PARA MAIS INFORMAÇÕES,VISITE WWW.P65WARNINGS.CA.GOV.
ATENÇÃO: desligue sempre a fontede alimentação CA da tomada antes
de desligar a fonte de alimentação
CA do concentrador de oxigénio.

180 - PT PN MN232-C4 L | Manual do usuárioATENÇÃO: não posicione o concentrador de oxigénio
de modo a dificultar o acesso ao cabo de alimentação.
ATENÇÃO: posicione sempre os tubos de administração
de oxigénio e os cabos de alimentação de forma a evitaro perigo de quedas ou estrangulamentos acidentais.
ATENÇÃO: ao utilizar o ocncentrador de oxigénio
num automóvel, barco ou em outras fontes CC
com a fonte de alimentação CC, certifique-se deque o veículo está ligado e em funcionamento
antes de ligar o concentrador de oxigénio. O não
seguimento destas instruções pode resultar no facto
de a fonte de alimentação não fornecer alimentação
ao concentrador de oxigénio.
ATENÇÃO: quando o automóvel no qual está a utilizaro concentrador de oxigénio é desligado, desligue
e remova o dispositivo do automóvel. Não guardeo concentrador de oxigénio num automóvel com uma
temperatura demasiado elevada ou demasiado baixa
ou noutros ambientes com temperaturas elevadas
ou baixas similares.
Nota: o equipamento de comunicação RF portátil
ou móvel pode afetar o equipamento elétrico médico.
ATENÇÃO: NÃO deixe o concentrador deoxigénio ou a fonte de alimentação ligados
ao veículo se a ignição estiver na posiçãoDESLIGADO. Fazê-lo pode descarregar
a bateria do veículo.
ATENÇÃO: o funcionamento ou armazenamento
do concentrador de oxigénio fora do respetivo
intervalo de temperatura de funcionamento normal
e intervalo de pressão ambiente (altitude) pode afetaro desempenho e reduzir o tempo de funcionamento
da bateria e/ou aumentar o tempo de carga da bateria.
Consulte os limites de temperatura de funcionamentoe armazenamento na secção Especificações deste
manual.
ATENÇÃO: para os concentradores de oxigénio
equipados com baterias: Armazene em local frescoe seco para ajudar a garantir uma maior vida útil dasua bateria. Armazenar o seu concentrador de oxigénio
durante longos períodos de tempo com temperaturas
elevadas, ou com uma bateria completamente carregada
ou descarregada, pode degradar a vida útil da bateria.
Não tente abrir a bateria: não existem peças reparáveis
no seu interior. Mantenha as baterias afastadas doalcance das crianças.
ATENÇÃO: UTILIZE APENAS as baterias fornecidas
pelo Fabricante. Para a eliminação adequada dabateria, entre em contacto com o seu Fornecedor do
Equipamento ou a sua agência governamental local
para se informar sobre os requisitos de eliminação.

FreeStyle Comfort
Nota: para atenuar os danos causados por uma falha
do concentrador ou da alimentação, o fornecedor do
equipamento deve avaliar a necessidade de oxigéniode reserva durante a instalação e periodicamente quando
estes atributos forem alterados:- O estado do doente- O ambiente em que o doente vive
- A capacidade de reabastecer o doente com
fornecimentos de reserva de oxigénio suplementar.
Nota: certifique-se de que a cânula está totalmente
inserida e fixa. Isto garante que o concentrador de oxigénio
pode detetar corretamente a inspiração para fornecimentode oxigénio. Durante a inalação, deverá ouvir ou sentir
o fluxo de oxigénio pelos tubos da cânula nasal.
Nota: as baterias de lítio podem perder a capacidade
permanentemente quando expostas a temperaturas
extremamente quentes com as baterias totalmentecarregadas ou completamente gastas. Para um
armazenamento prolongado, recomendamos que asbaterias sejam carregadas a um nível de 25 a 50%
e permaneçam num intervalo de temperatura de-20 °C – 20 °C (-4 °F – 68 °F).

Nota: se o concentrador de oxigénio tiver sido armazenado
durante um longo período de tempo fora das respetivas
condições ambientais normais de funcionamento
apresentadas na tabela de especificações, deve serpermitido que volte à temperatura de funcionamento
normal antes de ser ligado à alimentação.
Nota: o fornecedor do equipamento deve avaliar anecessidade de oxigénio de reserva no caso de falha
do concentrador ou da alimentação no momento da
instalação e periodicamente quando estes atributos
forem alterados:- O estado do doente- O ambiente em que o doente vive
- A capacidade de reabastecer o doente com
fornecimentos de reserva de oxigénio suplementar.

Nota: não opere o concentrador de oxigénio sem o filtro daentrada de ar colocado. Se for fornecido um segundo filtro,
introduza o filtro de substituição antes de limpar o filtro
sujo. Limpe o filtro sujo numa solução tépida de água esabão e depois seque muito bem antes de voltar a usar.
Nota: o fabricante não recomenda a esterilização deste
equipamento.
Nota: nunca existe o perigo de esgotar o oxigénio de uma
divisão ao usar o Concentrador de oxigénio.
AVISO: O CONCENTRADOR DE OXIGÉNIONUNCA DEVE SER UTILIZADO OU
LIGADO A UMA FONTE DE ALIMENTAÇÃOEXTERNA (CA/CC) SEM UMA BATERIAINSTALADA. NÃO TENTE ATIVAR A
UNIDADE NEM LIGÁ-LA À ALIMENTAÇÃOEXTERNA QUANDO A UNIDADE DA
BATERIA TIVER SIDO RETIRADA.
RETIRE A UNIDADE DA BATERIA APENAS QUANDOA UNIDADE ESTIVER DESATIVADA (OFF) E
DESLIGADA DA ALIMENTAÇÃO ELÉTRICA. NÃO
TENTE RETIRAR A UNIDADE DA BATERIA COM
A UNIDADE AINDA ATIVA (ON) OU LIGADA À FONTE
DE ALIMENTAÇÃO.

PN MN232-C4 L | Manual do usuário PT - 181

FreeStyle Comfort
Especificações
FreeStyle Comfort
Dimensões (com uma bateria individual)

25,4 x 18,5 x 7,9 cm (10,0 x 7,3 x 3,1 in)Dimensões (com uma bateria dupla)

27,9 x 18,5 x 7,9 cm (11,0 x 7,3 x 3,1 in)

Peso (com uma bateria individual)

2,3 kg (5 lbs)

Peso (com uma bateria dupla)

2,7 kg (6 lbs)

Nível sonoro nominal***39,93 dB(A) ± 0,42 dB(A) @ 2

Definições de fluxo

Definição pulsátil 1, 2, 3, 4, 5

Definição de saída de O2([@20bpm STPD] Todos os volumes +/- 15%)

Concentração de oxigénio*Monitor de oxigénio

Definição 1 - 210 ml/min
Definição 2 - 420 ml/min
Definição 3 - 630 ml/min
Definição 4 - 840 ml/min
Definição 5 - 1050 mL/min
90% (+5.5% / -3%)
Sim

Pressão máxima de saída<30 PSI

Sensibilidade de dosagem> -0,1 cm H2O (ajustável)Alimentação CA

100–240 V CA, 50–60 Hz, 120 W Máx.

Alimentação CC

11–18 V CC (10 Amp. máx.)

Tipo de bateriaIões de lítio

Capacidade da bateria
Tempo de recarga da bateria

Célula única: 6700 mAh; Célula dupla: 13 400 mAh
Bateria individual: 3,5 horas
Bateria dupla: 6 horas

Durações da unidade da bateria

Duração da bateria individual: Definição 2 - até 4 horas
Duração da bateria dupla: Definição 2 - até 8 horas

Temperatura de funcionamento**

5 °C a 40 °C (41 °F a 104 °F)

Temperatura de transporte e armazenamento(Unidade)-25 °C a +70 °C (-13 °F a +158 °F) até 3048 m (10 000 pés)Temperatura de transporte e armazenamento(Bateria)-25 °C a +70 °C (-13 °F a +158 °F) até 3048 m (10 000 pés)Altitude de funcionamento**-381 a 3048 m (-1250 a 10 000 pés) (testado a 700–1060 hPa)

Humidade de funcionamento

Humidade relativa de 15–95% (sem condensação)Humidade de armazenamento

0–90% Sem condensação

Garantia limitada

Consultar a Declaração de Garantia* Com base numa pressão atmosférica de 14,7 psi (101 kPa) a 21 °C (70 °F)** A unidade funcionará em todos os ambientes, desde que o ambiente esteja dentro dos intervalos de temperatura
e humidade especificados. A utilização fora destas especificações de funcionamento pode limitar a capacidade
do concentrador de satisfazer a especificação de Concentração de oxigénio a taxas de fluxo por litro mais altas.
O ecrã de aquecimento poderá continuar a ser apresentado até que o concentrador alcance a concentração alvo ou até
15 minutos. Continue a utilizar a unidade durante este período.
*** Nível sonoro medido de acordo com o método de ensaio n.º 14-1 10/2018 MDS-Hi.
A vida útil esperada do dispositivo é de, no mínimo, cinco anos.

182 - PT PN MN232-C4 L | Manual do usuário

FreeStyle Comfort
Instruções de utilização
1. Coloque e posicione o FreeStyle Comfort num local
adequado para a utilização.
a. O concentrador deve ser bem ventilado para que
todos os filtros de entrada de ar e saída de ar não
fiquem obstruídos.
b. O Concentrador deve ser posicionado de tal
modo que todos os indicadores sonoros e visuais
sejam facilmente vistos e ouvidos.
c. Certifique-se de que os filtros de entrada doar estão colocados antes de utilizar a unidade.
Se faltar um filtro na unidade, contacte
o fornecedor do seu equipamento.
2. Instale uma unidade da bateria no concentrador.
a. Alinhe a bateria com a estrutura inferior a partir
do centro do FreeStyle Comfort.

3. Introduza a fonte de alimentação CA ou CC
no recetáculo da porta de alimentação no cantoinferior direito frontal da unidade.

4. Ligue uma cânula nasal ao encaixe de saídade oxigénio. Certifique-se de que a ligação está
segura antes de utilizar.
NOTA: quaisquer cânulas utilizadas que não sejam
as recomendadas podem afetar o desempenho.

b. Deslize a bateria para dentro até ouvir um cliquepara confirmar que está encaixada na respetiva
posição.

Nota: para o fornecedor do equipamento: Sãorecomendados os seguintes acessórios de administração
de oxigénio para utilização com o FreeStyle Comfort:• Cânula nasal com tubo de 2,1 m (7 pés) (LPM máximo:
6 LPM): Número de peça CAIRE 5408-SEQ• Estão disponíveis informações adicionais sobre
os acessórios recomendados em www.caireinc.com.
5. Ligue o FreeStyle Comfort premindo o botão
de energia e selecione a sua definição de fluxo
de ar prescrita, utilizando os botões para aumentar(+) ou reduzir (-) o fluxo.

PN MN232-C4 L | Manual do usuário PT - 183

FreeStyle Comfort

É emitido um breve sinal sonoro para indicar queo dispositivo está ligado.
Nota: quando o FreeStyle Comfort é ligado inicialmente
a definição de fluxo predefinida é 2. Se a definição foralterada, a unidade liga na definição de fluxo anterior.
6. C oloque a cânula nasal no rosto, seguindo as instruções
do fabricante fornecidas com a cânula e respire
normalmente. Deverá sentir um sopro de oxigénio
administrado pela cânula nasal de cada vez que inala.
7. Para alterar a definição do fluxo, prima os botões
para aumentar (+) ou reduzir (-) o fluxo até obtera taxa de fluxo pretendida. É normal ouvir uma
diferença no som ao alterar as definições de fluxo.

Frequência respiratória
15
20
25
30
35
40

Definição de Volume de pulso sem
autoSAT (ml)*
1
2
3
4
5
14,0
10,5
8,4
7,0
6,0
5,3

28,0
21,0
16,8
14,0
12,0
10,5

42,0
31,5
25,2
21,0
18,0
15,8

56,0
42,0
33,6
28,0
24,0
21,0

70,0
52,5
42,0
35,0
30,0
26,3

8. Desligue o FreeStyle Comfort premindo o botão
de energia.
Nota: o ecrã do visor entra no modo de espera
e escurece após 20 segundos. Prima qualquer botão
para reativar o ecrã do modo de espera.
9. Para funcionalidades avançadas, consulte
as instruções de navegação do menu.

CARACTERÍSTICAS ESPECIAIS
DO PRODUTO
autoSAT®
A funcionalidade autoSAT patenteada da CAIRE
permite que o FreeStyle Comfort forneça um volume
consistente de oxigénio (bolus) por respiração, atéuma frequência respiratória máxima. À medida que
a frequência respiratória do utilizador aumenta duranteatividades como caminhar, a funcionalidade autoSAT
ajusta o funcionamento do dispositivo para manter um
volume de bolus definido.

Frequência respiratória
15
20
25
30
35
40

Definição de Volume de pulso autoSAT
(ml)*
1
2
3
4
5
10,5
10,5
10,5
10,5
10,5
10,5

21,0
21,0
21,0
21,0
21,0
21,0

31,5
31,5
31,5
31,5
30,0
26,3

42,0
42,0
42,0
35,0
30,0
26,3

*A tolerância é de +/- 15%

184 - PT PN MN232-C4 L | Manual do usuário

52,5
52,5
42,0
35,0
30,0
26,3

UltraSense®
Este dispositivo sensível de dosagem por pulsação
deteta as alterações de pressão durante a sua respiração
e administra a pulsação de oxigénio quando inspira.
autoDOSE®
Em caso de ocorrência de um alarme de “sem inspiração”,
esta funcionalidade efetua pulsações de oxigénio a
20 inspirações por minuto na atual definição de fluxo
da unidade até o estado de alarme ser resolvido.

FreeStyle Comfort
Fontes de alimentação
O FreeStyle Comfort pode ser alimentado de trêsformas:• Unidade da bateria recarregável• Alimentação CA• Alimentação CC
As informações sobre a operação de um destes
componentes estão disponíveis nas secções quese seguem.

REMOVER A BATERIA

Empurre o botão de libertação da bateria para baixo,
afastando-o do FreeStyle Comfort. MANTENDO estebotão premido, deslize a bateria para fora afastando-a
do concentrador.
AVISO: O CONCENTRADOR DE OXIGÉNIODEVE ESTAR DESATIVADO E DESLIGADODA ALIMENTAÇÃO EXTERNA ANTESDE RETIRAR A UNIDADE DA BATERIA.

AVISO: UTILIZE APENAS AS FONTES DE
ALIMENTAÇÃO DESCRITAS NESTE MANUAL DO UTILIZADOR. A UTILIZAÇÃO DE
FONTES DE ALIMENTAÇÃO E UNIDADESDA BATERIA QUE NÃO SEJAM DA CAIRE
PODE PROVOCAR DANOS NA UNIDADE E
UM FUNCIONAMENTO INCORRETO.

CONJUNTO DE BATERIAO FreeStyle Comfort pode ser alimentado pela unidadeda bateria recarregável fornecida com o aparelho.
Não existem limitações operacionais do concentrador
ao funcionar com a unidade da bateria. Todas as taxas
de fluxo estão disponíveis na alimentação atravésda bateria.

VERIFICAR A CARGA DA UNIDADEDA BATERIA

Para verificar a carga da unidade da bateria instalada
no FreeStyle Comfort, consulte o ícone da bateriae o tempo restante previsto no visor.

INSTALAR A BATERIA

Alinhe a bateria com a estrutura inferior a partir docentro do FreeStyle Comfort. Empurre a bateria para
dentro até ouvir um estalido para confirmar que ficou
encaixada no local.

80–100%60–80%

40–60%

20–40%

20%

PN MN232-C4 L | Manual do usuário PT - 185

FreeStyle Comfort
Nota: se a fonte de alimentação estiver ligada
ao dispositivo, é apresentado um indicador de ficha
em vez da percentagem de carga restante.
Para verificar a carga da bateria quando não está
instalada no FreeStyle Comfort, coloque-a plana e primao botão TEST. As luzes indicadoras do medidor dabateria (25–100%) acendem-se à direita do botão TEST
para indicarem o nível de carga da unidade da bateria.

CARREGAMENTO INICIAL

A bateria nova fornecida com o seu FreeStyle Comfort
só está parcialmente carregada quando é enviada defábrica. Antes da primeira utilização, deve carregar
completamente a unidade da bateria utilizando energia
CA ou CC.

INFORMAÇÕES GERAISDE CARREGAMENTO

O FreeStyle Comfort recarrega a unidade sempre que
esteja instalada no concentrador e o concentrador
esteja ligado a uma fonte de alimentação CA ou CC
externa (exceto numa aeronave).
A bateria está totalmente carregada quando apresentar
100%.
Alternativamente, a unidade da bateria pode sercarregada utilizando o carregador de bancada opcional.

186 - PT PN MN232-C4 L | Manual do usuário

TEMPOS DE FUNCIONAMENTO
DA BATERIA

Quando a alimentação externa CA e CC é desligada,o concentrador muda automaticamente para autilização da energia da bateria se estiver instalada uma
bateria carregada. O quadro abaixo mostra os tempos
habituais para uma unidade da bateria nova. Os tempos
da unidade da bateria mantêm-se constantes seja qual
for a velocidade da sua respiração.
Definição
PulsátilDuração dabateria individual(8 células)Duração dabateria dupla(16 células)1

até 8 horas

até 16 horas2

até 4 horas

até 8 horas3

até 3 horas

até 6 horas4

até 2,25 horas

até 4,5 horas5

até 2 horas

até 4 horas

TEMPO TÍPICO DE RECARREGAMENTO
DA BATERIAO período típico para recarregar completamente a
sua unidade da bateria totalmente descarregada é de
3,5 horas para uma bateria individual e 6 horas parauma bateria dupla.

Nota: a bateria pode não começar a carregar inicialmente
se a unidade da bateria estiver completamente gasta.

VIDA ÚTIL DA BATERIA

As baterias FreeStyle Comfort foram concebidas para
durarem 300 ciclos de carga/descarga.

FreeStyle Comfort
Fonte de alimentação CA e CC
O FreeStyle Comfort inclui uma fonte de alimentação
CA e CC de origem.
A fonte de alimentação CA inclui o seguinte:	a. Unidade de alimentação CA que funciona
entre 100-240 V, 50-60 Hz.
	b. Cabo de alimentação CA para estabelecer
ligação à unidade de alimentação e à tomada
de CA correspondente.

O transformador CC consiste num cabo que é ligado
diretamente à unidade e à tomada de CC.

ALIMENTAÇÃO CA

A alimentação CA destina-se à utilização em casa
ou onde se encontre alimentação de energia CApadrão disponível. O FreeStyle Comfort é totalmente
funcional com energia CA. O FreeStyle Comfort
também recarrega a unidade da bateria sempre que
exista energia CA disponível (exceto numa aeronave).
A unidade da bateria recarrega quer o FreeStyle
Comfort esteja ligado ou desligado.

2. Ligue o cabo de alimentação CA à unidadede alimentação CA.

AVISO: NÃO LIGUE A ALIMENTAÇÃOEXTERNA SEM UMA UNIDADEDA BATERIA INSTALADA NO
CONCENTRADOR DE OXIGÉNIO.
Para ligar o FreeStyle Comfort à alimentação CA:
1. Ligue a unidade de alimentação CA á entrada doconetor no canto inferior direito frontal do FreeStyle
Comfort. Certifique-se de que empurra para dentro
de modo a que a ligação fique segura.

3. Ligue a extremidade oposta do cabo de alimentação
CA à parede ou a outra tomada elétrica adequada.

PN MN232-C4 L | Manual do usuário PT - 187

FreeStyle Comfort
4. Ligue o FreeStyle Comfort premindo o botão deenergia e selecione a sua definição de fluxo de arprescrita, utilizando os botões para aumentar (+) ou
reduzir (-) o fluxo.

4. Ligue o FreeStyle Comfort premindo o botão deenergia e selecione a sua definição de fluxo de arprescrita, utilizando os botões para aumentar (+)
ou reduzir (-) o fluxo.
Nota: Se a unidade da bateria tiver de ser recarregada,o ícone da bateria no vidro apresenta uma única barra
cor de laranja.

ALIMENTAÇÃO CC

A alimentação CC destina-se à utilização em tomadaspara acessórios de automóveis, embarcações ou
outros veículos motorizados. O FreeStyle Comfort é
totalmente funcional com energia CC. O FreeStyle
Comfort também recarrega a unidade da bateria
se necessário sempre que exista alimentação CC
disponível. A unidade da bateria recarrega se a unidade
estiver ligada ou desligada, caso esteja em utilização
ou não.
AVISO: NÃO LIGUE A ALIMENTAÇÃOEXTERNA SEM UMA UNIDADEDA BATERIA INSTALADA NO
CONCENTRADOR DE OXIGÉNIO.
1. Ligue o veículo motorizado.
2. Ligue a fonte de alimentação CC ao recetáculo da
porta de alimentação no canto inferior direito frontal
do FreeStyle Comfort. Certifique-se de que empurrapara dentro de modo a que a ligação fique segura.

3. Ligue a extremidade oposta do cabo de alimentaçãoCC à saída de alimentação CC do automóvel ou deoutro veículo.
188 - PT PN MN232-C4 L | Manual do usuárioATENÇÃO: para evitar um possível incêndio
ou sobreaquecimento, certifique-se de que
a tomada de alimentação de 12 V CC estálivre de detritos e que a ficha do adaptador
da fonte de alimentação CC encaixa
corretamente e permanece firmemente
na respetiva posição.
ATENÇÃO: as superfícies metálicas da fonte dealimentação CC podem ficar MUITO QUENTES quandoestão a ser utilizadas devido à corrente elevada.
Deixar arrefecer as superfícies durante um períodode tempo suficiente antes de as tocar.
ATENÇÃO: não utilize a fonte de alimentação CC
com quaisquer extensões, divisores ou adaptadores.
A utilização de acessórios não aprovados podeprovocar o seu sobreaquecimento ou danificação.
ATENÇÃO: ligar o dispositivo a uma tomada de 12 V
CC pode esgotar a bateria do veículo se o motor não
estiver a funcionar, mesmo que o dispositivo esteja
desligado.
ATENÇÃO: não force o arranque um veículo enquantoa fonte de alimentação CC estiver ligada.

FreeStyle Comfort
Limpeza, Conservação
e Manutenção de Rotina
MANUTENÇÃO DE ROTINAManutenção preventiva
O dispositivo não necessita de manutenção. Os filtros
no interior do dispositivo destinam-se a durar toda
a vida útil do dispositivo. Se o dispositivo não
estiver a funcionar corretamente, a unidade deve serdevolvida ao fornecedor para reparação. A reparação
dos componentes internos no interior da caixa do
FreeStyle Comfort deve ser realizada pelo fornecedor
do oxigénio.
O operador da unidade é responsável pelo seguinte
procedimento de manutenção.
• Semanalmente – Limpar os filtros de entrada do ar
AVISO: NÃO TENTE ABRIR NEM REMOVER
AS TAMPAS DE PLÁSTICO DO CONCENTRADOR DE OXIGÉNIO. A UNIDADE SÓ
DEVE SER ABERTA POR UM TÉCNICO
DE ASSISTÊNCIA QUALIFICADO.

FILTROS DE ENTRADA DO AR

O ar entra no FreeStyle Comfort através de filtros
de entrada do ar situados na parte inferior do painel
frontal do concentrador de oxigénio. Estes filtros
impedem a entrada de pó e de outras partículas grandes
no interior da unidade.
No mínimo uma vez por semana, os filtros de entrada
do ar devem ser limpos. O seu fornecedor de oxigénio
poderá aconselhá-lo a limpá-los com maior frequência,
dependendo das condições de funcionamento. Sigaestes passos para limpar corretamente os filtros deentrada de ar.
1. Retire os dois filtros do painel frontal do FreeStyle
Comfort empurrando a patilha correspondente para
cima e puxando para fora.

AVISO: NÃO OPERE O CONCENTRADORDE OXIGÉNIO SEM QUE O FILTRO DE ADMISSÃO DE AR ESTEJA CORRETAMENTEPOSICIONADO.

CUIDADOS DA UNIDADE DA BATERIA

Não permita que a unidade da bateria seja expostaa líquidos.
Não tente desmontar ou abrir a unidade da bateria.
Não exponha a bateria a chamas.
Não deixe cair a unidade da bateria.
Mantenha a unidade da bateria afastada do alcancedas crianças.
Se a sua unidade da bateria parecer danificada, contacte
o seu fornecedor de oxigénio antes de utilizar.
Não coloque a bateria no interior ou perto de um forno
micro-ondas ou outros eletrodomésticos de cozinha.

ARMAZENAMENTO DA UNIDADEDA BATERIA

A bateria do concentrador de oxigénio não tem deestar completamente descarregada antes de recarregar.
É recomendada a recarga da bateria do concentradorde oxigénio após cada utilização.
Não deixe a bateria num local exposto a calor
excessivo, tal como exposto á luz solar direta ou nointerior de um automóvel. Fazê-lo poderá provocar
a deterioração do desempenho e da vida útil dabateria. Também poderá fazer com que a bateria gerecalor, fumo, incendiar-se ou explodir. A unidade dabateria deve ser armazenada num ambiente estável,caracterizado por um nível reduzido de humidade(inferior a 70% de HR) e livre de gases corrosivos.
Para evitar a formação de ferrugem, evite condições
que possam criar condensação, tais como flutuações
rápidas na temperatura ambiente.
Para um armazenamento durante um mês ou
mais, é recomendado um estado descarregado
ou parcialmente carregado entre 50% e 75%(dois a três LED). Depois de carregada, retire a bateria
do concentrador de oxigénio.

2. Lave ambos os filtros numa solução de água quente
e detergente.
3. Enxague cuidadosamente os filtros e remova oexcesso de água com uma toalha suave e sem cotão.
Certifique-se de que os filtros estão completamente
secos antes de voltar a instalá-los.
4. Instale os filtros limpos e secos.
PN MN232-C4 L | Manual do usuário PT - 189

FreeStyle Comfort
LIMPEZADESLIGUE a unidade e desconecte-a também daalimentação CA ou CC antes de efetuar qualquer
limpeza. NÃO pulverize diretamente a caixa exterior.
Utilize um pano ou esponja húmidos (não molhados).
Pulverize o pano ou a esponja com uma solução delimpeza doméstica suave para limpar a estrutura e as
fontes de alimentação. Proceda como indicado pelo
fabricante da solução de limpeza.
AVISO: DESLIGUE A UNIDADE EO CABO DE ALIMENTAÇÃO DA
TOMADA ELÉTRICA ANTES DE LIMPARA UNIDADE PARA EVITAR CHOQUESELÉTRICOS ACIDENTAIS E O PERIGO DE
QUEIMADURAS. APENAS O FORNECEDORDO SEU EQUIPAMENTO OU UM TÉCNICO
DE ASSISTÊNCIA QUALIFICADO DEVERÃOREMOVER AS TAMPAS OU REALIZARTAREFAS DE MANUTENÇÃO NA UNIDADE.
AVISO: DEVE SER EXERCIDO CUIDADO PARAEVITAR QUE O CONCENTRADOR DE OXIGÉNIO OU
A UNIDADE DA BATERIA SE MOLHEM OU A ENTRADA
DE SUBSTÂNCIAS LÍQUIDAS NO INTERIOR DA
UNIDADE. ISTO PODE PROVOCAR UMA AVARIA OU
PARALISAÇÃO E RESULTAR NUM MAIOR RISCO
DE CHOQUES ELÉTRICOS OU QUEIMADURAS.
AVISO: NÃO UTILIZE LÍQUIDOS DIRETAMENTE NA
UNIDADE. UMA LISTA DE AGENTES QUÍMICOSINDESEJÁVEIS INCLUI, MAS NÃO SE LIMITA AOSSEGUINTES: ÁLCOOL E PRODUTOS À BASE
DE ÁLCOOL, PRODUTOS À BASE DE CLORO
CONCENTRADOS (CLORETO DE ETILENO) E
PRODUTOS À BASE DE ÓLEO. ESTES PRODUTOS
NÃO DEVEM SER UTILIZADOS PARA LIMPAR A CAIXA
DE PLÁSTICO NO CONCENTRADOR, UMA VEZ QUE
PODEM DANIFICAR A UNIDADE.
AVISO: LIMPE A CAIXA, O PAINEL DE CONTROLOE O CABO DE ALIMENTAÇÃO COM UM DETERGENTEDOMÉSTICO SUAVE APLICADO COM UM PANOOU ESPONJA HÚMIDOS (NÃO MOLHADOS) E
DEPOIS SEQUE BEM TODAS AS SUPERFÍCIES. NÃO
PERMITA A ENTRADA DE LÍQUIDOS NO INTERIORDO CONCENTRADOR. PRESTE ESPECIAL ATENÇÃOÀ SAÍDA DE OXIGÉNIO PARA A LIGAÇÃO DA CÂNULA
A FIM DE GARANTIR QUE SE MANTÉM LIVRE DE PÓ,ÁGUA E PARTÍCULAS.

190 - PT PN MN232-C4 L | Manual do usuário

AVISO: USE UM PANO HÚMIDO (NÃO MOLHADO)OU ESPONJA PARA LIMPAR A UNIDADE DA BATERIA.
PRIMEIRO, PULVERIZE O PANO OU ESPONJA COM
UM PRODUTO DE LIMPEZA DOMÉSTICA SUAVE
E LIMPE A CAIXA DA UNIDADE DA BATERIA.
AVISO: NÃO EFETUE A LIMPEZA COM UM SISTEMADE LAVAGEM AUTOMATIZADO.

SUBSTITUIÇÃO DA CÂNULA

Siga sempre as instruções do fabricante da cânula
para uma utilização correta. Substitua a cânula
nasal ou os tubos de oxigénio a cada 3 a 6 meses(se utilizar diariamente, 3 meses) de acordo com asrecomendações do fabricante ou do fornecedor deoxigénio. O seu médico ou fornecedor de oxigénio
fornecer-lhe-á as instruções de limpeza e substituição.
Pode adquirir artigos de substituição adicionais através
do seu fornecedor de oxigénio.

FreeStyle Comfort
Condições de Alarme
N.º daprioridade

Descrição
do alarme

1

Avaria dosistema,Temperatura
elevada dabateria

Visor

Sinal
sonoroMais detalhes/Ação recomendada

1 bip acada 20
segundos

A unidade irá emitir um alarme quando atemperatura da bateria estiver a aproximar-se
do limiar de desativação de segurança.
Se o limiar de desativação for atingido, a unidade desativa-se. Se a alimentação externa
estiver ligada, é apresentado o ecrã de alarmee um sinal sonoro é emitido durante 2 min.
Efetue um teste com a bateria à temperatura
ambiente. Se o problema persistir, contacte
o fornecedor.

2

Avaria dosistema, Alta
temperatura

1 bip acada 20
segundos

A unidade emite um alarme quando atemperatura interna se aproximar do limiar
de desativação.
Se o limiar de desativação for atingido, aunidade desativa-se e é apresentado o ecrã dealarme e o sinal sonoro é emitido durante 2 min.
Certifique-se de que nenhuma saída deventilação da unidade está obstruída.
Se o problema persistir, contacte o fornecedor.

3

Avaria dosistema,Pressão
elevada(pressão
elevada dosistema ou
falha docompressor)

1 bip acada 20
segundos

4

Avaria dosistema,
Taxa de fluxo(Falha da
válvula OCD,
Obstrução datrajetória dogás)

1 bip acada 30
segundos

Se o problema persistir, contacte
o fornecedor.

5

Bateria fraca,
Desativação

1 bip acada 30
segundos

Emite um alarme até a bateria ficar
sem energia e a unidade se desativar.

Se o limiar de desativação for atingido, aunidade desativa-se e é apresentado o ecrã dealarme e o sinal sonoro é emitido durante 2 min.
Se o problema persistir, contacte o fornecedor.

Carregue a bateria.

PN MN232-C4 L | Manual do usuário PT - 191

FreeStyle Comfort
N.º daprioridade

Descrição
do alarme

6

Respiração
não detetada

Visor

Sinal
sonoroMais detalhes

1 bip acada 10minutosNão é detetada respiração durante
15 segundos.
Ativado após a primeira respiração ou quandoo ecrã de aquecimento é desativado, o que
ocorrer primeiro.
Verifique se a cânula apresenta dobras ou
fugas. Aumente a definição de sensibilidade.
Se o problema persistir, contacte
o fornecedor.

7

Aquecimento

nenhum

A unidade inicia a dosagem automática atéque seja detetada uma respiração ou atéque o Aquecimento termine, o que ocorrer
primeiro.

8

Baixa
concentração

1 bip acada 10minutos,até 30minutos

Se o problema persistir, contacte
o fornecedor.

9

Aviso debateria fraca

1 bip, semrepetição

Carregue a bateria.

10Frequência
respiratória
excedida

1 bip, semrepetição

A frequência respiratória excede os 40 bpm.

11

Avaria dosistema, Falha
da ventoinha

nenhumSe o problema persistir, contacte
o fornecedor.

12

Avaria dosistema,Baixa pressão

nenhumSe o problema persistir, contacte
o fornecedor.

13

Avaria dosistema, Baixa
temperatura

nenhumDeixe a unidade aquecer até à temperatura
ambiente. Se o problema persistir, contacte
o fornecedor.

192 - PT PN MN232-C4 L | Manual do usuário

Se a dosagem for acionada incorretamente,
reduza a definição da sensibilidade.
Se o problema persistir, contacte o fornecedor.

FreeStyle Comfort
N.º daprioridade

Descrição
do alarme

Informação

Arranque

Visor

Sinal
sonoroMais detalhes

1 bip deinformação

O ecrã é apresentado no arranque e deverá
ouvir um bip para confirmar que o sinal
sonoro está a funcionar.
O ecrã de arranque apresenta a versão
do firmware no canto inferior direito.

Informação

Ecrã defuncionamento
normal no
Modo autoSAT(sem alimentação CC
conetada)

nenhumInformação

Ecrã defuncionamento
normal com
autoSAT desativado (semalimentação
CC conetada)nenhumInformação

Definição
alterada

1 bip deinformação

Informação

Liga aalimentação
externa na
condição deequipamento
desligado

nenhumInformação

Liga aalimentação
externa na
condição defuncionamento

nenhumInformação

Desliga aalimentação
externa na
condição deequipamento
desligado

1 bip deinformação

Apresenta o ecrã CAIRE durante 2 segundos,
seguido pelo estado bateria e ligação durante
5 segundos antes de se desligar

Apresenta o ecrã CAIRE durante 2 segundose, em seguida, desliga-se.

PN MN232-C4 L | Manual do usuário PT - 193

FreeStyle Comfort
N.º daprioridade

Descrição
do alarme

Visor

Informação

Desliga aalimentação
externa na
condição defuncionamento(com a bateriapresente)

1 bip deinformação

Informação

No modo deespera - quando o botão
para cima ou
para baixo
é premido,
apresentao estado dabateria e decarregamento

nenhumInformação

Modo deavião comalimentação
da bateria

1 bip
longo

Informação

Modo de avião
enquanto está
ligado

1 bip
longo

194 - PT PN MN232-C4 L | Manual do usuário

SinalsonoroMais detalhes

O ecrã é apresentado durante 5 segundos e,em seguida, desliga-seFreeStyle Comfort
Instruções de navegação do menu
É possível entrar no Menu nos modos de Espera e deFuncionamento. Se ocorrer uma condição de alarme
no modo de funcionamento, a unidade sai do Menude forma automática. Se nenhum botão for premido
durante 20 segundos, a unidade sai do menu de forma
automática. No modo de Espera, poderá ter de premiro botão de Menu duas vezes para ativar a unidade e,em seguida, entrar no Menu.
Existem quatro ecrãs de Menu:
1. Ecrã de estado: apresenta o número da revisão dofirmware, o tempo de funcionamento, os ciclos dedescarga da bateria

Modo de voo

2. Modo de avião (LIGADO/DESLIGADO)
3. Sensibilidade: ajustável (1-5)
4. autoSAT (LIGADO/DESLIGADO)

Sensibilidade

Exemplo de ecrã de estado

UTILIZAÇÃO DO MENU DE DEFINIÇÕES
Para entrar no menu de Definições, prima o botão
Menu/Seleção
ou de Espera.

autoSATnos modos de Funcionamento

Para navegar através dos menus, prima.

Página de estado

PN MN232-C4 L | Manual do usuário PT - 195

FreeStyle ComfortPÁGINA DE ESTADO
A página de estado do FreeStyle Comfort mostra
a versão do firmware, as horas de funcionamentoe o número de ciclos da bateria ligada.

Para cancelar o Modo de voo, primaquandoo ícone CANCEL (CANCELAR) estiver visível.

Modo de voo ligado confirmado
Página de estado

Para desligar o Modo de voo, repita o procedimento
anterior mudando para OFF (DESLIGADO).

As horas de funcionamento são apresentadas em horas
inteiras.
As descargas da bateria são o número de ciclos em quea bateria ligada foi descarregada e recarregada.

Premirquando o ecrã apresenta o símbolo “Next”(Seguinte) fará a página do menu avançar para a opção
de página seguinte.

MODO DE VOOO Modo de voo destina-se a ser usado no Modo defuncionamento. O modo de voo fica automaticamente
inativo se a unidade for deixada em Modo de espera
quando alimentada através da bateria.

O FreeStyle Comfort é enviado com o Modo de voo
desligado.

Nota: Consulte a secção Ativar o modo de voo para obter
mais informações sobre o Modo de voo.

MODO DE SENSIBILIDADEEcrã principal do modo de voo

O FreeStyle Comfort é enviado com uma predefinição
do Modo de sensibilidade de 2 de fábrica e, em
condições normais, não deve necessitar de qualquer
ajuste. A definição de sensibilidade pode ser ajustada
entre os valores 1-5, sendo 1 a menos sensívele 5 a mais sensível.

Para ligar o modo, primaquando o ícone
CHANGE (ALTERAR) estiver visível. O valor mudará
para a palavra azul “ON” (LIGADO).

Ecrã do modo de sensibilidade com a Definição 2

Modo de voo ligado selecionado

Para confirmar o Modo de voo, primaquandoo ícone SAVE (GUARDAR) estiver visível.
O valor mudará para a palavra preta “ON” (LIGADO).

196 - PT PN MN232-C4 L | Manual do usuário

FreeStyle Comfort
Para ajustar o Modo de sensibilidade, primaquando o ícone CHANGE (ALTERAR) estiver visível.
O valor mudará para o número azul seguinte.

Definição 3 do modo de sensibilidade selecionada

Para confirmar a Definição 3 no Modo desensibilidade, primaquando o ícone SAVE(GUARDAR) estiver visível. O valor mudará parao “3” preto.

MODO AUTOSATO FreeStyle Comfort é fornecido com o Modo
autoSAT ligado.

Ecrã principal do modo autoSAT

Para desligar o modo, primaquando o ícone
CHANGE (ALTERAR) estiver visível. O valor mudará
para a palavra azul “OFF” (DESLIGADO).

Para cancelar a alteração do Modo de sensibilidade,prima
quando o ícone CANCEL (CANCELAR)
estiver visível.

Modo autoSAT desligado selecionado

Para confirmar o Modo autoSAT desligado, prima

Definição 3 do modo de sensibilidade confirmada

quando o ícone SAVE (GUARDAR) estivervisível. O valor mudará para a palavra preta “OFF”(DESLIGADO).
Para cancelar o Modo autoSAT ligado, primaquando o ícone CANCEL (CANCELAR) estivervisível.

Definição 5 (Máx) do modo de sensibilidade confirmada

Premirquando o ecrã apresenta o símbolo “Next”(Seguinte) fará a página do menu avançar para a opção
de página seguinte.

Modo autoSAT desligado confirmado

Para ligar o Modo autoSAT, repita o procedimento
anterior mudando para ON (LIGADO).
Premirquando o ecrã apresenta o símbolo “Next”(Seguinte) fará a página do menu avançar para a opção
de página seguinte.

PN MN232-C4 L | Manual do usuário PT - 197

FreeStyle Comfort
Resolução de Problemas
Problema

Causa provável

Solução

O FreeStyle Comfort apresentaum alarme e emite um bip
intermitente.

Consulte o quadro
de Condições de alarme.

Consulte o quadro de Condições de alarme.

O Concentrador de oxigénio
não liga.

A bateria está descarregada.

Alimente a unidade através da tomada de CA ou CC.

O concentrador não está ligado
à alimentação externa.

Assegure-se de que as conexões externas estão
bem efetuadas.

Avaria geral.

Execute um Ciclo de alimentação completoretirando a bateria e desligando qualquer fontede alimentação externa CA ou CC. Em seguida,
instale novamente a bateria e conclua o ciclode alimentação. Ligue novamente a fonte dealimentação externa, conforme necessário.
Contacte o fornecedor do equipamento e use outra
fonte de oxigénio, conforme necessário.

Restantes problemas.

Execute um Ciclo de alimentação completoretirando a bateria e desligando qualquer fontede alimentação externa CA ou CC. Em seguida,
instale novamente a bateria e conclua o ciclode alimentação. Ligue novamente a fonte dealimentação externa, conforme necessário.
Mude para outra fonte de oxigénio, conforme
disponível, e contacte o fornecedor do equipamento.

Nota: alguns alarmes fazem com que a unidade se desligue.
Nota: durante uma perda de energia total, o sistema de alarmes é reiniciado (elimina todos os alarmes anteriores)e só apresenta um alarme se a condição persistir depois de restaurada a alimentação.

198 - PT PN MN232-C4 L | Manual do usuário

FreeStyle Comfort
Informações de transporte
O FreeStyle Comfort cumpre todas as diretrizes
da FAA referentes à utilização a bordo de aviões
comerciais. Antes de qualquer viagem (de avião,comboio, cruzeiro, etc.), deverá:• Garantir que o concentrador se encontra em boas
condições de funcionamento.
• Contacte a empresa de transporte pelo menos
48 horas antes da partida para informar que irá viajar
com um FreeStyle Comfort.
É recomendado transportar sempre baterias extra
nas suas viagens. Para viagens aéreas, providencie
a autonomia da bateria suficiente para o tempo
em terra (check-in antes do voo, segurança, etc.)
e transbordos. A maioria das companhias aéreas exige
que tenha um número adequado de unidades de bateria
completamente carregadas para alimentar o dispositivo
durante, pelo menos, 150% da duração máxima da
viagem prevista (tempos de voo, tempo em terra antes
e após o voo, ligações e atrasos inesperados).

1. Ligue o POC utilizando APENAS a alimentação
da bateria quando não estiver ligada qualquer fontede alimentação externa.

2. Ative o Modo de voo usando um dos métodos
seguintes:
a. Consulte a secção de Instruções de navegação
do menu para ativar o Modo de voo.

Nota: nos voos, recomendamos que as baterias extra
sejam transportadas num saco de transporte protegido.

REGULAMENTO DA FAA

As baterias sobresselentes devem ser protegidas
individualmente para evitar curtos-circuitos através dacolocação na embalagem de comercialização originalou através de outra forma de isolamento dos terminais,
por exemplo, ao aplicar fita adesiva sobre os terminais
expostos ou colocar cada bateria num saco plásticoseparado ou numa bolsa protetora, e transportadas
apenas em bagagens de mão.

b. Prima e mantenha premidos os botões Para cimae Para baixo durante 5 segundos até ouvir um bip
longo e o símbolo de Avião for apresentado.

ATIVAR O MODO DE AVIÃOAlimentação da bateria
O POC pode ser utilizado numa aeronave de acordo com
os procedimentos de funcionamento normais utilizando
APENAS a alimentação da bateria quando não estiver
qualquer fonte de alimentação externa ligada

3. Quando o símbolo do avião for apresentado, ligue
o POC.

Alimentação Externa
Numa aeronave, se existir uma tomada elétrica
disponível, funcional e que a sua localização sejaadequada para utilizar como fonte de alimentação
do POC, poderá servir de reserva para as baterias do
POC. Neste caso, deve ser seguido o procedimento
recomendado relativamente à transição da bateria paraa alimentação elétrica da aeronave:
Nota: devido às limitações das aeronaves, as baterias
não carregam no Modo de avião.

PN MN232-C4 L | Manual do usuário PT - 199

FreeStyle Comfort
Acessórios

INSERIR A UNIDADE NO SACO DE
TRANSPORTE

Para um desempenho e segurança apropriados, use
apenas estes acessórios listados fornecidos pela CAIRE
através do seu fornecedor de oxigénio. A utilização
de acessórios não incluídos abaixo pode prejudicaro desempenho e/ou segurança do concentrador.

1. Introduza a unidade no fundo do saco, alinhandocorretamente os filtros e as aberturas da bateria.

Acessórios padrão do FreeStyle Comfort
Bolsa de transporte

MI396-1

MochilaMI459-1

Alça a tiracolo

MI397-1

Fonte de alimentação CA (todos oscabos incluídos)PW036-1SCabo de alimentação CC

CD041-1

Filtro de partículas grossas externo(direita)

FI226-1

Filtro de partículas grossas externo(esquerda)

FI227-1

2. Feche o fecho de correr em torno da bolsa.

Acessórios opcionais disponíveis
Unidade da bateria de 8 células
adicional

BT034-1S

Unidade da bateria de 16 células

BT035-1S

Carregador da unidade da bateria
de bancada

BT036-1

Bolsa de transporte para todos os
acessórios

MI372-2

Nota: podem estar disponíveis opções adicionais paracabos de alimentação específicos de cada país, talcomo mencionado anteriormente. Contacte a CAIRE
ou o seu fornecedor de oxigénio, caso seja necessário
encomendar opções alternativas.

Instruções do saco de transporte
Para inserir e fixar o seu FreeStyle Comfort no saco detransporte, siga as seguintes instruções.

200 - PT PN MN232-C4 L | Manual do usuário

INSTALAÇÃO DA ALÇA A TIRACOLO
1. A alça a tiracolo pode ser afixada diretamente
à unidade ou ao saco de transporte.
2. Prenda a alça no anel que se encontra na unidade
ou no saco.

FreeStyle Comfort
Eliminação e Reutilização
Devolva sempre o FreeStyle Comfort, incluindo todos
os componentes, ao seu fornecedor do equipamentopara uma eliminação adequada. Pode também entrar
em contracto com as entidades locais para obter
instruções acerca da correta eliminação da bateria.
A unidade pode ser reutilizada entre pessoas, masdeve ser devolvida ao fornecedor do equipamentopara reprocessamento.

REEE e RoHS
Este símbolo destina-se a recordar os
proprietários do equipamento de que devem entregá-lo
a uma instalação de reciclagem no final da sua vidaútil, de acordo com a Diretiva relativa aos Resíduos
de Equipamentos Elétricos e Eletrónicos (REEE).

Materiais em contacto direto com
o operador• Invólucro do concentrador: PC/ABS• Painel de controlo do concentrador: Poliéster EBG7
ou equivalente• Saída de oxigénio, Bocal: Alumínio• Rótulo da unidade: Lexan• Conectores do cabo: Policarbonato/cloreto de vinilo• Cabo de alimentação: PVC, Metal• Fonte de alimentação: Lexan 940 (Policarbonato)• Unidade da bateria: PC/ABS• Unidade da bateria, etiquetas da fontede alimentação: Película de poliéster
• Estojo de transporte do concentrador: Cordura,
neopreno, forro em nylon• Alça: Polipropileno, acetil, vinilo, Cordura, neopreno

Os nossos produtos estão em conformidade com adiretiva de Restrição de Substâncias Perigosas (RoHS).
Não contêm mais do que quantidades residuais dechumbo ou de outros materiais perigosos.
ATENÇÃO: para uma eliminação adequada,
entre em contacto com o fornecedor do seu
equipamento ou agência governamental
local para obter as instruções de eliminação.
ATENÇÃO: elimine sempre a peneira molecular em
conformidade com todos os regulamentos de controlo
ambiental federais, estatais e locais.

PN MN232-C4 L | Manual do usuário PT - 201

FreeStyle Comfort
Testes de compatibilidade eletromagnética (EMC)
Orientações e Declaração do Fabricante­—Emissões eletromagnéticas
O FreeStyle Comfort destina-se a ser usado no ambiente eletromagnético especificado abaixo. O cliente ou utilizador
do FreeStyle Comfort deve certificar-se de que este é usado no ambiente descrito.
Teste de Emissões
Emissões RF
CISPR 11
Emissões RF
CISPR 11
Emissões harmónicas
IEC 61000-3-2
Flutuações de tensão/emissões inconstantes
IEC 61000-3-3

Conformidade

Ambiente eletromagnético - Orientações

Grupo 1O FreeStyle Comfort utiliza energia de RF apenas para o seu
funcionamento interno. Por conseguinte, as suas emissões deRF são muito baixas e não é provável que causem interferências nos
equipamentos eletrónicos próximos.

Classe B
Classe A
Em conformidade

202 - PT PN MN232-C4 L | Manual do usuário

A utilização do FreeStyle Comfort é adequada em todos osestabelecimentos, incluindo os estabelecimentos domésticos e os que
se encontram ligados à rede pública de fornecimento de energia debaixa voltagem que abastece os edifícios usados para fins domésticos.

FreeStyle Comfort
Orientações e Declaração do Fabricante—Imunidade eletromagnética

O FreeStyle Comfort destina-se a ser utilizado no ambiente eletromagnético especificado abaixo. O cliente ou o utilizador
do FreeStyle Comfort deve garantir que é utilizado em tal ambiente.
Teste de imunidade Nível de teste IEC 60601 Nível de conformidade Ambiente eletromagnético – orientações
O piso deve ser de madeira, cimento
Ambiente
ou azulejo. Se os pavimentos estiverem
Contacto ±8 kV
Contacto ±8 kV
eletromagnético –cobertos com um material sintético,orientações
Descarga de ar ±15 kV
Descarga de ar ±15 kV
a humidade relativa deve ser de,
IEC 61000-4-2
pelo menos, 30%.
±2 kV para linhas deA qualidade da alimentação elétrica±2 kV para linhas defornecimento energético
Transiente/disparo
fornecimento energético deve ser equivalente à de instalações
rápido elétrico
de cuidados de saúde profissionais típicas
e de ambientes de cuidados de saúde
IEC 61000-4-4±1 kV para linhas deN/Ddomésticos.
entrada/saída
Linha(s) a linha(s) ±1 kV

Sobretensão
IEC 61000-4-5±2 kV modo comum em
linhas de CA±1 kV diferencial em
linhas de CA

Linha(s) para linha(s)±1 kV

A qualidade da alimentação elétrica
deve ser equivalente à de instalações
de cuidados de saúde profissionais típicas
e de ambientes de cuidados de saúde
domésticos.

±2 kV modo comum em
linhas de E/S no exterior
0% UT para 0,5 ciclo
0% UT para 0,5 ciclo(0°, 45°, 90°, 135°, 180°, (0°, 45°, 90°, 135°, 180°,
225°, 270°, 315°)
225°, 270°, 315°)
A qualidade da alimentação elétrica
deve ser equivalente à de instalações
Quedas de tensão,de cuidados de saúde profissionais típicas
interrupções curtas e 0% U para 1 ciclo (0°)
0%
U
para
1
ciclo
(0°)
T
T
e de ambientes de cuidados de saúde
variações de tensão
domésticos. Se o utilizador do FreeStylenas linhas de entrada
70% UT (30% submerso
70% UT (30% submerso Comfort necessitar de um funcionamentode fornecimento
em UT) para 25/30 ciclos em UT) para 25/30 ciclos continuado durante interrupções dede energia
energia, recomendamos que o FreeStyle
(0°)
(0°)
IEC 61000-4-11
Comfort seja alimentado por uma fonte dealimentação ininterrupta ou bateria.
0% UT para 250/300
0% UT para 250/300
ciclos (0°)
ciclos (0°)
Os campos magnéticos de frequência
Campo magnético
de energia devem estar a um nível
de frequência decaracterístico de um local típico em
3A/m
3A/m
energia (50/60 Hz)
ambientes de cuidados de saúde
IEC 61000-4-8
profissionais e domésticos.
NOTA UT é a tensão de CA antes da aplicação do nível de teste.

NOTA Em alguns casos, o FreeStyle Comfort poderá reiniciar após uma descarga eletrostática significativa e poderá
necessitar de um Ciclo de alimentação completo para retomar o funcionamento normal. Consulte a secção Resolução
de problemas para obter mais detalhes.

PN MN232-C4 L | Manual do usuário PT - 203

FreeStyle Comfort
Orientações e Declaração do Fabricante—Imunidade eletromagnética

O FreeStyle Comfort destina-se a ser utilizado no ambiente eletromagnético especificado abaixo. O cliente ou o utilizador
do FreeStyle Comfort deve certificar-se de que este é utilizado no ambiente descrito.
Teste deNível de teste IEC 60601 Nível de conformidade
Ambiente eletromagnético – orientações
imunidade
RF Conduzida
10 Vrms
10 Vrms
O equipamento de comunicações por
RF portátil e móvel não deve ser utilizado
IEC 61000-4-6
150 kHz a 80 MHz
a uma distância inferior de qualquer peçado FreeStyle Comfort, incluindo cabos, doque a distância de separação recomendadacalculada a partir da equação aplicável
RF irradiada
10 V/m
10 V/mà frequência do transmissor.
IEC 61000-4-3
80 MHz a 2700 MHz
Distância de separação recomendadad = 1.2 P

d = 1.2 P = 80 MHz a 800 MHz
d = 1.2 P = 800 MHz a 2700 MHz
em que P é a potência nominal máxima
de saída do transmissor em watts(W) de acordo com o fabricante dotransmissor e d é a distância de separação
recomendada em metros (m).
As forças de campo de transmissores deRF fixos, conforme determinado por uma
pesquisa eletromagnética do local,a não
deve ser inferior ao nível de conformidade
em cada intervalo de frequência.b
Poderão ocorrer interferências nas
imediações de equipamentos assinalados
com o seguinte símbolo:

NOTA 1 A 80 MHz e 800 MHz, aplica-se o intervalo de frequência superior.
NOTA 2 Estas diretrizes podem não se aplicar em todas as situações. A propagação eletromagnética é afetada pela
absorção e reflexão de estruturas, objetos e pessoas.
a
As forças de campo de transmissores fixos, como as estações de base para telefones de rádio (telemóveis/sem fios)e rádios móveis terrestres, rádio amador, transmissão de rádio AM e FM e transmissão de televisão não se podem
prever teoricamente com precisão. Para avaliar o ambiente eletromagnético para transmissores de RF fixos, deverá
ser considerado um estudo eletromagnético do local. Se a força do campo medido no local onde o FreeStyle Comfort
é utilizado exceder o nível de conformidade RF aplicável indicado acima, o FreeStyle Comfort deve ser observado para
determinar o seu funcionamento normal. Se for observado um desempenho anormal, poderão ser necessárias medidas
adicionais, tais como reorientar ou relocalizar o FreeStyle Comfort.
b

Ao longo do intervalo de frequência de 150 kHz a 80 MHz, as forças de campo devem ser inferiores a 10 V/m.

204 - PT PN MN232-C4 L | Manual do usuário

FreeStyle Comfort
Distâncias de separação recomendadas entre os equipamentos de comunicação deRF portáteis e móveis e o FreeStyle Comfort

O FreeStyle Comfort destina-se a ser utilizado num ambiente eletromagnético no qual as perturbações de RF irradiadas
sejam controladas. O cliente ou o utilizador do FreeStyle Comfort pode ajudar a prevenir interferências eletromagnéticas
mantendo uma distância mínima entre o equipamento de comunicações por RF portátil e móvel (transmissores)
e o FreeStyle Comfort conforme recomendado abaixo, de acordo com a potência máxima de saída do equipamentode comunicações.
Potência nominal
Distância de separação de acordo com a frequência do transmissor
máxima de saídamdo transmissor
150 kHz a 80 MHz
80 MHz a 800 MHz
800 MHz a 2,5 GHz= 1.2 P= 1.2 Pd = 1.2 P
d
d
W
0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
Para transmissores com uma potência nominal de saída máxima não indicada acima, a distância de separação
recomendada (d) em metros (m) pode ser estimada utilizando a equação aplicável à frequência do transmissor,
em que P é a potência nominal máxima de saída do transmissor em Watts (W) segundo o fabricante do transmissor.
NOTA 1 A 80 MHz e 800 MHz, aplica-se a distância de separação para o intervalo de frequência superior.
NOTA 2 Estas diretrizes podem não se aplicar em todas as situações. A propagação eletromagnética é afetada pela
absorção e reflexão de estruturas, objetos e pessoas.

Classificação

• CISPR 11/EN 55011 Classe B Grupo 1,	"Equipamento Industrial, Científico e Médico (ISM)"• FCC Parte 15, Subparte B – Classe B
	Radiadores não intencionais

Grau de proteção contra choques elétricos:
Equipamento do tipo BF
que fornece um determinado		grau de proteção contra choques elétricos		em relação a		1) corrente de fuga admissível;		2) fiabilidade da ligação à terra de proteção
		(se presente).
		Não se destina a aplicação cardíaca direta.

IP22 - Proteção da entrada de dedos ou objetos semelhantes com mais de 12,5 mm de diâmetro. A águade gotejamento vertical não deve ter qualquer efeito
nocivo quando a caixa estiver inclinada a um ângulo
de 15° em relação à sua posição normal.

Tipo de proteção contra choques elétricos:
Classe II	A proteção contra choques elétricos é			conseguida através de um isolamento duplo.

Testes independentes para a Norma para equipamentos
médicos elétricos:
	Testado para cumprir a• I EC 60601-1 edição 3.1: Equipamento médico
elétrico – Parte 1: Requisitos gerais de segurança.
• CAN/CSA C22.2 N.º 60601-1-8 edição 2.1,
Equipamento médico elétrico – Parte 1: Requisitos gerais de segurança

Método de limpeza e controlo de infeções permitido:
Consulte a secção "Limpeza, Conservação
e Manutenção Adequada" do Manual do utilizador
deste FreeStyle.
Grau de segurança da aplicação na presença de gases
anestésicos inflamáveis:	Equipamento não apropriado para tal aplicação.
Modo de funcionamento: Funcionamento contínuo.

Proteção de potenciais interferências eletromagnéticas
ou outras entre o equipamento e outros dispositivos.
	• Testado para cumprir a EN 60601-1-2/IEC 60601-1-2: 4.ª edição.
	• Testado para cumprir a RTCA/DO160
	Secção 21 Categoria M.

PN MN232-C4 L | Manual do usuário PT - 205

FreeStyle Comfort
Gebruikersbediening en indicatoren van de systeemstatus
ISO 7000
Gebruikershandleiding;gebruiksinstructies. Reg.nr. 1641

ISO 7010: Grafische symbolen – Veiligheidskleuren
en veiligheidssymbolen – Geregistreerde
veiligheidssymbolen

Het land en de datum van fabricage.
De ‘CC’ geeft de tweeletterige landcode
van het land van fabricage aan.
De fabricagedatum is in de opmaak
JJJJ-MM-DD. Reg.nr. 6049

Het lezen van de instructiehandleiding
is verplicht. Reg.nr. M002
Uit de buurt van open vuur,
vlammen en vonken bewaren.
Open ontstekingsbronnen en roken
verboden. Reg.nr. P003

Droog en uit de buurt van regenbewaren. Reg.nr. 0626

Rook niet in de buurt van het apparaat oftijdens de bediening ervan. Reg.nr. P002

Stapellimiet op aantal, waarbij ‘n’ het
maximaal toegestane aantal eenheden
aangeeft. Reg.nr. 2403
Catalogusnummer. Reg.nr. 2493

Waarschuwing. Reg.nr. W001
IEC 60417

Serienummer. Reg.nr. 2498

Toegepast onderdeel van type BF(mate van bescherming tegen elektrischeschokken). Reg.nr. 5333

Bereik opslag- of bedrijfstemperatuur.
Reg.nr. 0632

Apparatuur van klasse II, dubbelgeïsoleerd. Reg.nr. 5172

Vochtigheidsbereik opslag. Reg.nr. 2620
Bereik atmosferische druk. Reg.nr. 2621
Naam en adres van de fabrikant.
Reg.nr. 3082
Deze kant omhoog. Reg.nr. 0623

EU-verordening 2017/745
Gevolmachtigde vertegenwoordiger
in de Europese Gemeenschap
Als op de unieke
hulpmiddelenidentificatie (UDI) van
het product het CE####-symbool staat,
voldoet het apparaat aan de vereisten
betreffende medische hulpmiddelen.
Het CE####-symbool geeft het nummer
van de aangemelde instantie aan.
Aanvullende symbolen
Uit de buurt bewaren van ontvlambare
materialen, olie en vet.

Breekbaar, voorzichtig. Reg.nr. 0621

Niet demonteren.
Bevat gevaarlijke stoffen. Reg.nr. 3723
Debiet verhogen.
Importeur. Reg.nr. 3725
Debiet verlagen.
O2-uitlaat

Menuknop.
Stroomschakelaar. Reg.nr. 5009
RTCA/DO160 Deel 21
conformcategorie M.

206 - NL PN MN232-C4 L | Gebruikershandleiding

De fabrikant van deze POC heeftbepaald dat dit apparaat voldoet aan alle
toepasselijke FAA-acceptatiecriteria
en aan het gebruik aan boord vanvliegtuigen.

FreeStyle Comfort
De fabrikant van deze POC heeftbepaald dat dit apparaat voldoet aan alle
toepasselijke FAA-acceptatiecriteria
en aan het gebruik aan boord vanvliegtuigen.

Richtlijn 2012/19/EU van de Raad
AEEA
Dit symbool herinnert gebruikers eraanom de apparatuur aan het einde van delevensduur naar een recyclingbedrijf tebrengen, conform de AEEA-richtlijn(afgedankte elektrische en elektronische
apparatuur). Onze producten voldoen
aan de beperkingen van de richtlijn
inzake gevaarlijke stoffen (RoHS).
Ze bevatten slechts sporen van loodof andere gevaarlijke stoffen.

Recyclingsymbool
Gemachtigde vertegenwoordiger inZwitserland (ADD)
Als het apparaat is voorzien van het
UKCA-merkteken zoals aangegeven
met UKCA####, dat het nummer van
de aangemelde instantie aangeeft,
voldoet dit hulpmiddel aan de
UKCA-voorschriften.
21 CFR 801.15
Dit product mag op grond van de federale
wetgeving uitsluitend worden verkocht
door of op voorschrift van een arts.

IEC 60601-1-8
Technisch alarm met lage prioriteit
ISO 15223-1
Medisch hulpmiddel. Reg.nr. 5.7.7

ETL-certificering
Gecertificeerd voor zowel deAmerikaanse als de Canadese markt,
overeenkomstig de toepasselijke
Amerikaanse en Canadese normen.
Bevat FCC ID: WAP2001
FCC-kennisgeving
Dit apparaat omvat mogelijk CYBLE-022001-00,
waaronder de antenne 2450AT18B100 van Johnson
Technology, en voldoet aan deel 15 van de FCCregels. Het apparaat voldoet aan de vereisten voor
een modulaire zender, goedgekeurd zoals vermeld
in de openbare FCC-kennisgeving DA00-1407.
De werking van de zender is onderworpen aan devolgende twee voorwaarden: (1) Dit apparaat mag
geen schadelijke interferentie veroorzaken; en(2) Dit apparaat moet alle ontvangen interferentie
aanvaarden, waaronder interferentie die een
ongewenste werking kan veroorzaken. Op dit
product kunnen een of meer Amerikaanse eninternationale octrooien van toepassing zijn. Bezoek
onze website voor een lijst met toepasselijke
octrooien. Pat.: www.caireinc.com/corporate/patents

Uniek identificatienr. apparaat 5.7.10
A S/NZS 4417.1, AS/NZS 4417.2
Markering voor naleving van ACMAregelgeving – geeft aan dat de ACMAregelgeving wordt nageleefd (ADD)

Draadloze technologie
Verbindingstypen

SSP, iAP2, GATT

Frequentie

2402 tot 2480 MHz

Max RF-uitgangsvermogen+4 dBm

Werkbereik

10 m (klasse 2)

We adviseren u het apparaat tijdens het gebruik
op minimaal 1,1 cm (0,43") afstand van het lichaam
te houden. Niet van toepassing voor maskers,
luchtslangen en accessoires.

Grafische symbolen voor gebruik op het apparaat
IEC 60417
Gelijkstroom. Reg.nr. 5031
Wisselstroom. Reg.nr. 5032
IEC 60601-1
Druppelbestendig

PN MN232-C4 L | Gebruikershandleiding NL - 207

FreeStyle Comfort
Uw FreeStyle Comfort uitpakken
Controleer of alle hieronder
opgesomde onderdelen inde verpakking zitten. Neem
onmiddellijk contact op
met de leverancier van uw
zuurstofapparatuur als er
onderdelen ontbreken.

Standaardaccessoires• Draagbare zuurstofconcentrator• Draagtas• Schouderriem• 8-cellige accu
• Wisselstroomvoeding• Wisselstroomsnoer• Gelijkstroomvoeding• Gebruikershandleiding

Optionele accessoires• Extra 8-cellige accu
• 16-cellige accu
• Oplaadstation• Ruime accessoiretas

Uw FreeStyle Comfort leren kennen
Maak u vertrouwd met de belangrijkste functies van de FreeStyle Comfort en de gebruikersinterface.

Vooraanzicht
Riembevestiging

Gebruikersinterface

Zuurstofuitlaat

Luchtuitlaat

Luchtinlaat/-filters
Luchtinlaat/-filters

Houderlipje
voedingspoort

Accu
Ontgrendelingsknop
accu
208 - NL PN MN232-C4 L | Gebruikershandleiding

FreeStyle Comfort
Bovenaanzicht
Weergavescherm
Debiet verhogen

Aan/uit-knop

Menuknop

Debiet verlagen

Weergavescherm op accu
Debietinstelling

Accuweergave

autoSAT-indicator
Alarmweergave

Resterend
accupercentage

Bolustoedieningsindicator

Weergavescherm op wissel-/gelijkstroom

Debietinstelling
Accuweergave

Alarmweergaveruimte

Laadindicator

PN MN232-C4 L | Gebruikershandleiding NL - 209

FreeStyle Comfort
Wat is de zuurstofconcentrator?
De lucht die we inademen bevat ongeveer 21%
zuurstof, 78% stikstof en 1% overige gassen. Bij de
FreeStyle® Comfort® wordt omgevingslucht via deluchtinlaten in het apparaat gezogen. Deze lucht
stroomt vervolgens door een adsorberend materiaal,
een ‘moleculaire zeef’ genaamd. Dit materiaal scheidt
de zuurstof van de stikstof en laat alleen de zuurstof
door. Bijgevolg wordt hoogzuivere zuurstof naar degebruiker gevoerd.
De FreeStyle Comfort combineert een geavanceerde
zuurstofconcentrator en zuurstofbesparende
technologieën om een draagbaar apparaat te ontwikkelen
met het oog op optimale draagbaarheid en mobilisatie.
Dankzij de geavanceerde toediening met pulsdebiet
merkt het apparaat al snel wanneer de gebruiker inademt
en geeft het aan het begin van elke inademing een
zuurstofpuls vrij. De toediening met pulsdebiet is ideaal
voor activiteiten en tijd weg van huis. In tegenstelling
tot een continu debiet wordt er zuurstof bespaard en
wordt de accuduur verlengd. De FreeStyle Comfort
is licht genoeg om met een schouderriem te worden
gedragen. Het apparaat werkt op wisselstroom (muur),gelijkstroom (voertuig) of een accu.

Beoogd gebruik
De CAIRE FreeStyle Comfort-zuurstofconcentrator
is bedoeld voor de toediening van extra zuurstof.
Het apparaat is niet bedoeld ter ondersteuning of
instandhouding van de levensfuncties en biedt ook
geen voorzieningen voor het bewaken van patiënten.

Waarom heeft uw arts zuurstof
voorgeschreven?
Heel wat mensen lijden aan verschillende hart-, longen andere luchtwegaandoeningen. Een groot deelhiervan heeft baat bij de toediening van extra zuurstof
thuis, tijdens het reizen of bij dagelijkse activiteiten
weg van huis.
Zuurstof is een gas dat 21% van de omgevingslucht
vormt die we inademen en onze lichamen zijn
afhankelijk van een constante toevoer om goed tekunnen functioneren. Uw arts heeft u een debiet
of instelling voorgeschreven om uw specifieke
luchtwegaandoening te behandelen.

210 - NL PN MN232-C4 L | Gebruikershandleiding

Hoewel zuurstof een niet-verslavend geneesmiddel
is, kan ongeoorloofde zuurstoftoediening gevaarlijk
zijn. U moet medisch advies inwinnen voordat u dezezuurstofconcentrator gebruikt. De leverancier die u
de zuurstofapparatuur verstrekt, zal u tonen hoe u het
voorgeschreven debiet kunt instellen.

Gebruikersprofiel
Concentrators zijn bedoeld voor de toediening van extra
zuurstof aan gebruikers die lijden aan aandoeningen
die hun longen verhinderen op een efficiënte manier
zuurstof vanuit de lucht naar hun bloedbaan te voeren.
Draagbare zuurstofconcentrators (‘portable oxygen
concentrators’ of POC's) bewaren of bevatten geenzuurstof. Ze hoeven niet te worden bijgevuld en
kunnen op om het even welke plaats met wisselstroomof gelijkstroom worden opgeladen (behalve in een
vliegtuig). Dit biedt de gebruiker een relatieve
zelfstandigheid bij thuisgebruik, mobilisatie (zowel
in als van huis), mobiliteit en algemene levensstijl.
Voor het gebruik van een zuurstofconcentrator is een
voorschrift van een arts vereist.
De typische gebruiker van zuurstoftherapie
is ouder dan 65 jaar en lijdt aan verschillendeluchtwegaandoeningen, zoals chronische obstructieve
longaandoening (COPD). Gebruikers hebben
doorgaans goede cognitieve capaciteiten en moeten
hun ongemak kenbaar kunnen maken. Als degebruiker zijn ongemak niet kenbaar kan maken of deetikettering en gebruiksinstructies van de concentrator
niet kan lezen of begrijpen, dan is het gebruik ervanslechts aanbevolen onder toezicht van iemand diehier wel toe in staat is. Als er tijdens het gebruik van
de concentrator ongemak wordt ervaren, worden
gebruikers aanbevolen contact op te nemen met hun
zorgverlener. Ook worden gebruikers aangeraden overreservezuurstof te beschikken (d.w.z. zuurstofflessen)in het geval van een stroomonderbreking of een defectaan de concentrator. Er zijn geen andere specifieke
vaardigheden of gebruikerscapaciteiten vereist voor het
gebruik van de concentrator.

FreeStyle Comfort
Belangrijk!
Veiligheidsinstructies zijn als volgt vermeld:
WAARSCHUWING: BELANGRIJKE
VEILIGHEIDSINFORMATIE VOORGEVAREN DIE ERNSTIG LETSELKUNNEN VEROORZAKEN.
LET OP: Belangrijke informatie om schadeaan de FreeStyle Comfort te voorkomen.
Opmerking: Deze informatie vereist extra aandacht.

Indicaties voor gebruik
WAARSCHUWING: HET IS ERG
BELANGRIJK DAT ALLEEN HETVOORGESCHREVEN ZUURSTOFNIVEAUWORDT GEKOZEN. WIJZIG DE
DEBIETSELECTIE NIET TENZIJ DITWERD AANGEGEVEN DOOR EEN
BEVOEGDE CLINICUS.
WAARSCHUWING: DIT HULPMIDDEL IS NIETGETEST OF GOEDGEKEURD VOOR GEBRUIKBIJ PEDIATRISCHE PATIËNTEN.
WAARSCHUWING: VOLGENS DE AMERIKAANSEFEDERALE WETGEVING MAG DIT APPARAATALLEEN WORDEN VERKOCHT OF VERHUURD
OP VOORSCHRIFT VAN EEN ARTS OF EEN ANDEREERKENDE PROFESSIONELE ZORGVERLENER.
WAARSCHUWING: DIT APPARAATMAG NIET WORDEN GEBRUIKT VOORLEVENSONDERSTEUNING. EXTRA TOEZICHT KANNODIG ZIJN VOOR EEN GEBRUIKER DIE ONGEMAKTIJDENS HET GEBRUIK VAN HET APPARAATNIET KENBAAR KAN MAKEN. GEBRUIKERS MET
AUDITIEVE EN/OF VISUELE BEPERKING(EN)VEREISEN MOGELIJK HULP OP HET VLAK VAN
ALARMBEWAKING.
WAARSCHUWING: VOOR ELKE GEBRUIKER DIENENINDIVIDUELE INSTELLINGEN VOOR PULSDOSERING TE
WORDEN BEPAALD OVEREENKOMSTIG DE BEHOEFTEIN RUST, IN BEWEGING EN TIJDENS REIZEN.
WAARSCHUWING: DIT APPARAAT IS BEDOELDVOOR GEBRUIK BIJ THUISZORG EN SOORTGELIJKEPOLIKLINISCHE OMGEVINGEN.

WAARSCHUWING: HET KAN ZIJN DAT SOMMIGEADEMHALINGSINSPANNINGEN VAN DE PATIËNT DECONSERVERINGSFUNCTIE NIET AANSTUREN.
WAARSCHUWING: DE FREESTYLE COMFORT IS NIETBEDOELD VOOR GEBRUIK BIJ EEN PATIËNT DIE EEN
TRACHEOTOMIE HEEFT ONDERGAAN.
WAARSCHUWING: ZWANGERE OF BORSTVOEDENDEVROUWEN MOGEN GEEN ACCESSOIRESGEBRUIKEN DIE WORDEN AANBEVOLEN IN DEZE
HANDLEIDING, ZE KUNNEN FTALATEN BEVATTEN.

Contra-indicaties voor gebruik
WAARSCHUWING: IN BEPAALDEOMSTANDIGHEDEN KAN GEBRUIK VAN
NIET-VOORGESCHREVEN ZUURSTOFGEVAARLIJK ZIJN. DIT APPARAAT IS
ALLEEN BESTEMD VOOR GEBRUIKOP VOORSCHRIFT VAN EEN ARTS.
WAARSCHUWING: DIT APPARAAT IS NIETBEDOELD VOOR GEBRUIK IN OF IN DE BUURT VAN
OMGEVINGEN MET MAGNETISCHE RESONANTIE,ZOALS EEN MRI-SCANNER.
WAARSCHUWING: NIET BESTEMD VOORGEBRUIK IN DE NABIJHEID VAN ONTVLAMBARE
ANESTHETICA.
WAARSCHUWING: ZOALS BIJ ELK ELEKTRISCH
APPARAAT KRIJGT DE GEBRUIKER MOGELIJK TE
MAKEN MET ONDERBREKINGEN VAN DE WERKINGTEN GEVOLGE VAN EEN STROOMSTORING OF DENOODZAAK OM DE ZUURSTOFCONCENTRATORDOOR EEN ERKENDE TECHNICUS TE LATENNAZIEN. DE ZUURSTOFCONCENTRATOR IS NIET
AANGEWEZEN VOOR GEBRUIKERS VAN WIE DE
GEZONDHEID NADELIG ZOU WORDEN BEÏNVLOEDIN GEVAL VAN EEN TIJDELIJKE ONDERBREKING.
WAARSCHUWING: DIT APPARAAT IS NIET
GOEDGEKEURD VOOR GEBRUIK IN ALLEOMGEVINGEN. BLOOTSTELLING AAN BEPAALDEELEKTROMAGNETISCHE BRONNEN KAN LEIDENTOT INTERFERENTIE OF ONBEDOELDE PRESTATIESVAN HET APPARAAT. IN GEVAL VAN INTERFERENTIE
MOET HET APPARAAT ONMIDDELLIJK WORDENUITGESCHAKELD.

WAARSCHUWING: DIT APPARAAT IS BEDOELDVOOR GEBRUIK DOOR DE PATIËNT DIEZUURSTOFTHERAPIE KRIJGT.

PN MN232-C4 L | Gebruikershandleiding NL - 211

FreeStyle Comfort
Veiligheidsrichtlijnen
WAARSCHUWING: ER MOGEN GEENWIJZIGINGEN WORDEN AANGEBRACHTAAN DEZE APPARATUUR.
WAARSCHUWING: DIT APPARAAT LEVERTHOOGZUIVERE ZUURSTOF DIE EEN SNELLEVERBRANDING BEVORDERT. ZORG ERVOOR DATMEN NIET ROOKT EN DAT ER ZICH GEEN OPEN
VLAMMEN BEVINDEN IN DEZELFDE RUIMTE ALS:
(1) DIT APPARAAT OF (2) ZUURSTOFBEVATTENDEACCESSOIRES. NIET-NALEVING VAN DEZEWAARSCHUWING KAN LEIDEN TOT ERNSTIGE
BRAND, MATERIËLE SCHADE EN/OF FYSIEK LETSELOF DE DOOD.
WAARSCHUWING: GEBRUIK HET APPARAAT NIET IN
EEN AFGESLOTEN OF BESLOTEN RUIMTE (BIJV. EEN
KLEINE KOFFER OF HANDTAS) MET ONVOLDOENDE
VENTILATIE. DIT KAN OVERVERHITTING VAN DEZUURSTOFCONCENTRATOR VEROORZAKEN EN DE
GOEDE WERKING ERVAN BEÏNVLOEDEN.
WAARSCHUWING: DE CONCENTRATOR MOETWORDEN GEPLAATST OP EEN LOCATIE ZONDER
ROOK, VERVUILENDE STOFFEN OF DAMPEN.
WAARSCHUWING: ALS ZICH EEN ERNSTIGINCIDENT VOORDOET MET DIT APPARAAT, MOET DEGEBRUIKER HET INCIDENT ONMIDDELLIJK MELDENAAN DE LEVERANCIER EN/OF DE FABRIKANT.
EEN ERNSTIG INCIDENT WORDT GEDEFINIEERD
ALS LETSEL, DE DOOD OF DE KANS OP LETSEL/
DE DOOD ALS HET INCIDENT ZICH OPNIEUWVOORDOET. DE GEBRUIKER KAN HET INCIDENTOOK MELDEN BIJ DE BEVOEGDE AUTORITEIT IN HETLAND WAAR HET INCIDENT PLAATSVOND.
WAARSCHUWING: BEPAALDEZUURSTOFTOEDIENINGSACCESSOIRESDIE NIET ZIJN GENOEMD VOORGEBRUIK IN COMBINATIE MET
DEZE ZUURSTOFCONCENTRATOR,KUNNEN DE WERKING VAN HETAPPARAAT NADELIG BEÏNVLOEDEN.
DE AANBEVOLEN ACCESSOIRES STAANVERMELD IN DEZE HANDLEIDING.
WAARSCHUWING: ALS DE ZUURSTOFCONCENTRATORGEVALLEN IS, BESCHADIGD IS OF BLOOTGESTELDIS AAN WATER, NEEM DAN CONTACT OP MET UWLEVERANCIER VOOR EEN INSPECTIE EN EVENTUELE
REPARATIE VAN HET APPARAAT. GEBRUIK DEZUURSTOFCONCENTRATOR NIET ALS HET SNOER
OF DE STEKKER IS BESCHADIGD.
WAARSCHUWING: VOORKOM SCHADE OF LETSELTIJDENS VERVOER IN EEN VOERTUIG DOOR DEZUURSTOFCONCENTRATOR STEVIG TE BEVESTIGEN
MET EEN GORDEL OF ANDERSZINS.

212 - NL PN MN232-C4 L | Gebruikershandleiding

WAARSCHUWING: DE REGELGEVING VAN HETAMERIKAANSE DEPARTMENT OF TRANSPORTATION(DOT, HET MINISTERIE VAN VERVOER) EN DE
VERENIGDE NATIES (VN) EISEN DAT DE ACCUUIT HET APPARAAT WORDT VERWIJDERD
TIJDENS INTERNATIONALE VLUCHTEN ALSDE ZUURSTOFCONCENTRATOR WORDTINGECHECKT ALS BAGAGE. WANNEER DEZUURSTOFCONCENTRATOR WORDT VERVOERD,
MOET DE ACCU OOK VERWIJDERD EN CORRECTVERPAKT WORDEN.
WAARSCHUWING: LEES DEZE HANDLEIDING EN
GA NA OF U DE INHOUD BEGRIJPT VOORDATU DE APPARATUUR IN GEBRUIK NEEMT. ALS U
DE WAARSCHUWINGEN EN INSTRUCTIES NIETBEGRIJPT, DIENT U CONTACT OP TE NEMEN MET DE
LEVERANCIER VAN UW APPARATUUR ALVORENSU DEZE IN GEBRUIK NEEMT; DIT OM LETSELOF SCHADE TE VERMIJDEN.
WAARSCHUWING: VRAAG ONMIDDELLIJKMEDISCHE HULP ALS U ONGEMAK OF EENMEDISCHE NOODTOESTAND ERVAART.
WAARSCHUWING: GEBRUIK DE ZUURSTOFCONCENTRATOR NIET BIJ TEMPERATUREN DIE BUITEN DE
BEDRIJFSSPECIFICATIES LIGGEN. DIT KAN HET VERMOGEN VAN HET APPARAAT BEÏNVLOEDEN OM DEGESPECIFICEERDE ZUURSTOFCONCENTRATIE TE
BEREIKEN. RAADPLEEG HET DEEL ‘SPECIFICATIES’IN DEZE HANDLEIDING VOOR MEER INFORMATIE
OVER HET TEMPERATUURBEREIK.
WAARSCHUWING: GEBRUIK GEEN OLIE, SMEER,
PRODUCTEN OP BASIS VAN PETROLEUM
OF ANDERE ONTVLAMBARE PRODUCTENIN COMBINATIE MET ZUURSTOFHOUDENDE
ACCESSOIRES OF DE ZUURSTOFCONCENTRATOR.
GEBRUIK ALLEEN LOTIONS OF ZALVEN OP BASIS
VAN WATER DIE COMPATIBEL ZIJN MET ZUURSTOF.
ZUURSTOF VERSNELT DE VERBRANDING VAN
ONTVLAMBARE STOFFEN.
WAARSCHUWING: DOOR INCORRECT GEBRUIK VAN
DE ACCU KAN DEZE VERHITTEN, VLAM VATTENEN ERNSTIG LETSEL VEROORZAKEN. DOORBOOR
DE ACCU NIET, KLOP ER NIET OP, LAAT HEM NIET
VALLEN, STAP ER NIET OP EN STEL DE ACCU NIET
BLOOT AAN STERKE STOTEN OF SCHOKKEN.
HET GEBRUIK VAN EEN BESCHADIGDE ACCU KANLEIDEN TOT PERSOONLIJK LETSEL.
WAARSCHUWING: VEROORZAAK GEEN
KORTSLUITING VAN DE METALENCONTACTKLEMMEN VAN DE ACCU MET METALENVOORWERPEN, ZOALS SLEUTELS OF MUNTEN.
DIT KAN VONKEN OF OVERMATIGE HITTEVEROORZAKEN.

FreeStyle Comfort
WAARSCHUWING: ALS U DE DRAAGBAREZUURSTOFCONCENTRATOR BUITEN GEBRUIKTMET WISSELSTROOMVOEDING, MAG DE VOEDINGALLEEN AANGESLOTEN WORDEN OP EEN
AARDLEKSCHAKELAAR.
WAARSCHUWING: DE ZUURSTOFCONCENTRATORMAG NIET WORDEN GEBRUIKT NAAST OF GESTAPELD
OP ANDERE APPARATUUR. ALS AANPALEND OFGESTAPELD GEBRUIK ONVERMIJDELIJK IS, MOETWORDEN GECONTROLEERD OF HET APPARAATNORMAAL WERKT.
WAARSCHUWING: GEBRUIK VAN NIETGESPECIFICEERDE KABELS EN ADAPTERS, METUITZONDERING VAN KABELS EN ADAPTERS DIEWORDEN VERKOCHT DOOR DE FABRIKANT VAN
DE MEDISCHE ELEKTRISCHE APPARATUUR ALSVERVANGENDE ONDERDELEN VOOR INTERNECOMPONENTEN, KAN RESULTEREN IN EEN
VERHOOGDE UITSTOOT OF VERMINDERDEIMMUNITEIT VAN DE ZUURSTOFCONCENTRATOR.
WAARSCHUWING: GEBRUIK ALLEEN DE SPANNING
DIE WORDT VERMELD OP HET TYPEPLAATJE VAN
HET APPARAAT.
WAARSCHUWING: GEBRUIK GEEN VERLENGSNOEREN IN COMBINATIE MET DIT APPARAAT EN BEPERKHET AANTAL APPARATEN DAT IS AANGESLOTENOP HETZELFDE STOPCONTACT. HET GEBRUIK VAN
VERLENGSNOEREN KAN DE PRESTATIES VAN HETAPPARAAT NEGATIEF BEÏNVLOEDEN. INDIEN ER
TOCH TE VEEL APPARATEN OP EEN STOPCONTACTWORDEN AANGESLOTEN, DAN BESTAAT HET RISICOOP OVERBELASTING WAARDOOR DE ZEKERINGEN
KUNNEN SPRINGEN. ALS DE ZEKERINGEN NIET GOED
WERKEN, KAN DIT ZELFS BRAND VEROORZAKEN.
LET OP: Raadpleeg de probleemoplossingstabel in deze handleiding in geval
van een alarm of als u vaststelt dat dezuurstofconcentrator niet naar behoren
werkt. Neem contact op met de leverancier
van uw apparatuur als u er niet in slaagtom het probleem op te lossen.
LET OP: Om te voorkomen dat de garantie zijn
geldigheid verliest, dient u alle aanwijzingen vande fabrikant in acht te nemen.

WAARSCHUWING: VOER GEENONDERHOUD UIT DAT GEEN DEELUITMAAKT VAN DE MOGELIJKEOPLOSSINGEN IN DEZE HANDLEIDING.
WAARSCHUWING: GEBRUIK ALLEEN ACCESSOIRESDIE DOOR DE FABRIKANT ZIJN AANBEVOLEN.
GEBRUIK VAN ANDERE ACCESSOIRES KANGEVAARLIJK ZIJN, ERNSTIGE SCHADETOEBRENGEN AAN UW ZUURSTOFCONCENTRATOREN DE GARANTIE ANNULEREN.
WAARSCHUWING: DE FABRIKANT RAADT AAN OM
EEN ALTERNATIEVE BRON VAN EXTRA ZUURSTOFTE GEBRUIKEN IN GEVAL VAN EEN STROOMUITVAL,
ALARMSITUATIES OF MECHANISCHE STORINGEN.
RAADPLEEG UW ARTS OF DE LEVERANCIER VAN
UW APPARATUUR VOOR HET VEREISTE SOORTRESERVESYSTEEM.
WAARSCHUWING: BLOKKEER NOOIT DE
OPENINGEN VAN DE LUCHTINLAAT/-UITLAAT. STEEKGEEN VOORWERPEN IN DE OPENINGEN VAN HETAPPARAAT EN VOORKOM DAT ER VOORWERPENIN VALLEN. DIT KAN OVERVERHITTING VAN DEZUURSTOFCONCENTRATOR VEROORZAKEN EN DE
GOEDE WERKING ERVAN BEÏNVLOEDEN.
WAARSCHUWING: ROKEN TIJDENS GEBRUIKVAN ZUURSTOF IS DE VOORNAAMSTEOORZAAK VAN BRANDWONDEN EN DAARMEEGEPAARD GAAND OVERLIJDEN. VOLG DEZE
VEILIGHEIDSWAARSCHUWINGEN:WAARSCHUWING: STA NIET TOE DAT ER WORDTGEROOKT, DAT ER KAARSEN BRANDEN OF OPEN
VUUR IS IN DE KAMER WAAR HET APPARAAT OF
DE ZUURSTOFHOUDENDE ACCESSOIRES ZICH
BEVINDEN.
WAARSCHUWING: ROKEN TERWIJL U EENZUURSTOFCANULE DRAAGT, KAN LEIDEN TOT
BRANDWONDEN IN HET GELAAT OF ZELFS DE DOOD.
WAARSCHUWING: ALS U ROOKT, KUNNEN
DE VOLGENDE 3 STAPPEN UW LEVEN REDDEN:
SCHAKEL DE ZUURSTOFCONCENTRATOR UIT,VERWIJDER DE CANULE EN VERLAAT DE KAMERWAAR HET APPARAAT ZICH BEVINDT.

PN MN232-C4 L | Gebruikershandleiding NL - 213

FreeStyle Comfort
WAARSCHUWING: PLAATS DUIDELIJKE BORDEN
MET ‘ROKEN VERBODEN – GEBRUIK VAN
ZUURSTOF’ IN DE WONING OF ELDERS WAARZUURSTOF WORDT GEBRUIKT. GEBRUIKERSEN HUN ZORGVERLENERS MOETEN INGELICHTWORDEN OVER DE GEVAREN DIE ROKEN IN
DE AANWEZIGHEID VAN OF BIJ GEBRUIK VAN
MEDISCHE ZUURSTOF INHOUDT.
WAARSCHUWING: GEBRUIK UWZUURSTOFCONCENTRATOR NIET IN DE
AANWEZIGHEID VAN ONTVLAMBARE GASSEN.
DIT KAN SNEL BRAND VEROORZAKEN EN LEIDEN
TOT MATERIËLE SCHADE, LICHAMELIJK LETSELOF OVERLIJDEN.
WAARSCHUWING: ALS U DE CANULE AFNEEMTEN PLAATST OP KLEDING, BEDLINNEN, EEN SOFAOF KUSSENS ONTSTAAT ER EEN STEEKVLAM BIJ
BLOOTSTELLING AAN EEN SIGARET, WARMTEBRONOF VLAM.
WAARSCHUWING: PLAATS GEEN NEUSCANULE OP
OF ONDER KLEDING, BEDLINNEN OF KUSSENS VAN
EEN STOEL. ALS HET APPARAAT IS INGESCHAKELD
MAAR NIET WORDT GEBRUIKT, ZAL DE ZUURSTOF HET
MATERIAAL ONTVLAMBAAR MAKEN. CONTROLEER OF
DE ZUURSTOFCONCENTRATOR IS UITGESCHAKELDWANNEER HIJ NIET WORDT GEBRUIKT.
WAARSCHUWING: GOEDGEKEURDE ACCESSOIRESVOOR ZUURSTOFTOEDIENING ZIJN ONTWORPENOM BRANDVERSPREIDING TE VERMINDEREN VOORDE VEILIGHEID VAN DE PATIËNT. HET GEBRUIK VAN
NIET-GOEDGEKEURDE ACCESSOIRES KAN HETRISICO OP LETSEL OF SCHADE ALS GEVOLG VAN
BRAND VERHOGEN.
WAARSCHUWING: DE ZUURSTOFCONCENTRATOR
MOET UIT STAAN EN ZIJN LOSGEKOPPELDVAN EXTERNE VOEDING VOORDAT U DE ACCUVERWIJDERT.
WAARSCHUWING: KLEINEONDERDELEN KUNNEN MOGELIJKVERSTIKKINGSGEVAAR OPLEVEREN.
WAARSCHUWING: VERMIJD CONTACT MET DEONTLUCHTINGSOPENING GEDURENDE LANGERDAN 10 SECONDEN, WANT DEZE KAN WARM ZIJN.
WAARSCHUWING: WIND OF STERKELUCHTSTROMEN KUNNEN EEN NEGATIEVEINVLOED HEBBEN OP DE JUISTE TOEDIENINGVAN ZUURSTOFTHERAPIE.
WAARSCHUWING: DIT PRODUCT KAN UBLOOTSTELLEN AAN CHEMISCHE STOFFEN ZOALSNIKKEL, WAARVAN DE STAAT CALIFORNIË WEETDAT HET KANKERVERWEKKEND IS. GA VOOR MEER
INFORMATIE NAAR WWW.P65WARNINGS.CA.GOV.

214 - NL PN MN232-C4 L | Gebruikershandleiding

LET OP: Koppel de wisselstroomvoeding
altijd los van het contact voor u dewisselstroomvoeding loskoppelt van
de zuurstofconcentrator.
LET OP: Plaats de zuurstofconcentrator nooit zodanig
dat het snoer moeilijk toegankelijk is.
LET OP: Plaats de zuurstoftoevoerslangen en
netsnoeren altijd zodanig dat struikelgevaar en het
gevaar van onopzettelijke wurging worden voorkomen.
LET OP: Bij gebruik van de zuurstofconcentrator in een
auto, boot of met andere gelijkstroombronnen met degelijkstroomvoeding, moet het voertuig zijn opgestarten draaien voordat u de zuurstofconcentrator aansluit.
Als deze instructies niet nageleefd worden, is het
mogelijk dat de voeding geen stroom levert aan dezuurstofconcentrator.
LET OP: Als de auto waarin u de zuurstofconcentrator
gebruikt uitgeschakeld is, moet het apparaat worden
losgekoppeld en uit de auto worden verwijderd.
Berg de zuurstofconcentrator niet op in een heel
warme of koude auto of andere vergelijkbare
omgevingen met een hoge of lage temperatuur.
Opmerking: Draagbare en mobiele RF-communicatieapparatuur kan elektrische medische apparatuurbeïnvloeden.
LET OP: Laat de zuurstofconcentrator ofde voedingsbron NIET aangesloten op het
voertuig als de ontsteking in de uit-standstaat. Dit kan de batterij van het voertuig
ontladen.
LET OP: Als de zuurstofconcentrator buiten het normale
bedrijfstemperatuurbereik en omgevingsdrukbereik(hoogte) werkt of wordt opgeslagen, kan dit deprestaties beïnvloeden en de zelfstandige gebruiksduur
van de accu verminderen en/of de oplaadduur vande accu verlengen. Raadpleeg het deel Specificaties
in deze handleiding voor meer informatie over het
temperatuurbereik voor opslag en gebruik.
LET OP: Voor zuurstofconcentrators met accu's:
Bewaar de accu op een koele en droge plaats om
de levensduur ervan te verzekeren. Langdurig
opbergen van de zuurstofconcentrator bij hoge
temperaturen of met een volledig geladen of ontladenaccu, kan de levensduur van de accu beperken.
Probeer de accu niet te openen. De accu bevat geen
onderdelen die door de gebruiker kunnen worden
onderhouden. Houd accu's buiten het bereik van
kinderen.
LET OP: GEBRUIK ALLEEN batterijen die afkomstig
zijn van de fabrikant. Neem contact op met deleverancier van uw apparatuur of met de bevoegde
lokale autoriteiten voor de afdankingsrichtlijnen voor
een correcte afdanking van accu's.

FreeStyle Comfort
Opmerking: Om schade veroorzaakt door de concentratorof als gevolg van een stroomstoring te beperken, moet deleverancier van de apparatuur beoordelen of er behoefte
is aan back-upzuurstof tijdens de installatie en periodiek
wanneer deze kenmerken veranderen:

Opmerking: Als de zuurstofconcentrator voor lange tijd buiten
de normale bedrijfsomstandigheden zoals weergegeven inde tabel met specificaties is opgeslagen, moet het apparaat
eerst weer de normale bedrijfstemperatuur bereiken voordat
het wordt ingeschakeld.

- De toestand van de patiënt.

Opmerking: De leverancier van de apparatuur moet
beoordelen of er behoefte is aan back-upzuurstof bij een
concentrator- of stroomstoring bij installatie en periodiek
wanneer deze kenmerken veranderen:- De omgeving waarin de patiënt leeft.
- De mogelijkheid om de patiënt opnieuw van extra
zuurstof te voorzien.
Opmerking: Controleer of de canule correct is geplaatst
en goed vastzit. Zo kan de zuurstofconcentrator correct
de ademhaling detecteren voor de toediening van zuurstof.
Tijdens het inademen, moet u de zuurstoftoediening via
de takken van de neuscanule horen of voelen.

- De toestand van de patiënt.
- De omgeving waarin de patiënt leeft.
- De mogelijkheid om de patiënt opnieuw van extra
zuurstof te voorzien.

Opmerking: Lithiumbatterijen dreigen permanent hun
capaciteit te verliezen als ze worden blootgesteld aan
zeer hete temperaturen en volledig opgeladen of volledig
ontladen zijn. Voor langdurige opslag is het aanbevolen
de batterijen 25 tot 50% op te laden en binnen een
temperatuurbereik van -20°C–20°C (-4°F–68°F)
te bewaren.
Opmerking: Gebruik de zuurstofconcentrator niet als het
luchtinlaatfilter niet is bevestigd. Als een tweede filter ismeegeleverd, plaatst u het vervangingsfilter voordat u het
vervuilde filter reinigt. Maak het vuile filter schoon in een
warme oplossing van zeep en water en laat het volledig
drogen vóór gebruik.
Opmerking: De fabrikant raadt niet aan om dezeapparatuur te steriliseren.
Opmerking: Er is nooit risico op uitputting van zuurstof
in een ruimte waar u uw zuurstofconcentrator gebruikt.
WAARSCHUWING: DEZUURSTOFCONCENTRATOR MAG NIETWORDEN GEBRUIKT OF OP EEN EXTERNE(GELIJKSTROOM-/WISSELSTROOM-)VOEDING WORDEN AANGESLOTEN
ALVORENS DE ACCU IS GEPLAATST.
PROBEER HET APPARAAT NIET IN TESCHAKELEN OF OP EEN EXTERNEVOEDING AAN TE SLUITEN ZONDER DAT
DE ACCU WERD GEPLAATST.
VERWIJDER DE ACCU PAS WANNEER HETAPPARAAT IS UITGESCHAKELD EN VAN DE
VOEDING IS LOSGEKOPPELD. PROBEER DE ACCUNIET TE VERWIJDEREN ZOLANG HET APPARAATNOG IS INGESCHAKELD OF AAN DE VOEDING ISGEKOPPELD.

PN MN232-C4 L | Gebruikershandleiding NL - 215

FreeStyle Comfort
Specificaties
FreeStyle Comfort
Afmetingen (met enkele accu)

10,0 x 7,3 x 3,1 inch (25,4 x 18,5 x 7,9 cm)

Afmetingen (met dubbele accu)

11,0 x 7,3 x 3,1 inch (27,9 x 18,5 x 7,9 cm)

Gewicht (met enkele accu)

5 lbs (2,3 kg)

Gewicht (met dubbele accu)

6 lbs (2,7 kg)

Nominaal geluidsniveau***

39,93 dB(A) ± 0,42 dB(A) bij 2

Flowinstellingen

Pulsinstelling 1, 2, 3, 4, 5O2-uitvoerinstelling

Instelling 1 - 210 ml/min
Instelling 2 - 420 ml/min
Instelling 3 - 630 ml/min
Instelling 4 - 840 ml/min
Instelling 5 - 1050 ml/min([bij 20 bpm STPD] Alle volumes +/- 15%)

Zuurstofconcentratie*
Zuurstofmonitor

90% (+5,5% / -3%)
Ja

Maximale uitvoerdruk< 30 PSI

Doseergevoeligheid> -0,1 cm H2O (aanpasbaar)

Wisselstroom

100–240 VAC, 50–60 Hz, 120 W Max

Gelijkstroom

11–18 VDC (10 A max)Batterijtype

Lithium-ion

Batterijcapaciteit
Oplaadtijd accu
AccuduurBedrijfstemperatuur*

Enkele cel: 6700 mAh; Dubbele cel: 13400 mAh
Enkele accu: 3,5 uur
Dubbele accu: 6,0 uur
Duur enkele accu: Instelling 2 – max. 4 uur
Duur dubbele accu: Instelling 2 – max. 8 uur
5°C tot 40°C (41°F tot 104°F)

Vervoers- en opslagtemperatuur (apparaat)-25°C tot +70°C (-13°F tot +158°F) tot 3048 m (10.000 ft)

Vervoers- en opslagtemperatuur (accu)-25°C tot +70°C (-13°F tot +158°F) tot 3048 m (10.000 ft)

Gebruikshoogte**-381 tot 3048 m (-1250 tot 10.000 ft) (getest bij 700–1060 hPa)

Bedrijfsvochtigheid

15–95% relatieve vochtigheid (niet-condenserend)

Opslagvochtigheid

0% tot 90% niet condenserend

Beperkte garantie

Zie Garantieverklaring* Op basis van een luchtdruk van 101 kPa (14,7 psi) bij 21°C (70°F)** Het apparaat werkt in alle omgevingen zolang de omgeving voldoet aan de gespecificeerde temperatuur- en
vochtigheidswaarden. Bij gebruik buiten deze bedrijfsspecificaties dreigt het vermogen van de concentrator om aande zuurstofconcentratiespecificaties te voldoen te worden beperkt bij een hoger debiet. Het opwarmscherm kan blijven
weergeven totdat de concentrator de doelconcentratie of maximaal 15 minuten heeft bereikt. Blijf het apparaat gedurendedeze tijd gebruiken.
*** Geluidsniveau gemeten met testmethode nr. 14-1 10/2018 MDS-Hi.
De verwachte levensduur van dit apparaat bedraagt minimaal vijf jaar.

216 - NL PN MN232-C4 L | Gebruikershandleiding

FreeStyle Comfort
Gebruiksaanwijzing
1. Plaats en installeer de FreeStyle Comfort op een
gepaste gebruikslocatie.
a. De concentrator moet voldoende geventileerdzijn, zodat de luchtinlaatfilters en luchtuitlaat
niet worden geblokkeerd.
b. De concentrator moet zodanig worden geplaatst
dat alle hoorbare en zichtbare controlelampjes
en alarmen snel kunnen worden gezien engehoord.
c. Controleer of de luchtinlaatfilters zijn geplaatst
alvorens het apparaat te gebruiken. Als er een
filter ontbreekt op het apparaat moet u contact
opnemen met de leverancier van uw apparatuur.
2. Plaats een accu in de concentrator.
a. Lijn de accu uit met de onderste behuizing
vanuit het midden van de FreeStyle Comfort.

b. Schuif de accu naar binnen totdat u een klik
hoort om de vergrendeling aan te geven.

3. Steek de wissel- of gelijkstroomvoeding in devoedingspoort aan de voorkant, rechts onderaan
op het apparaat.

4. Sluit een neuscanule aan op dezuurstofuitlaataansluiting. Controleer
de aansluiting vóór gebruik.
OPMERKING: Ander dan het aanbevolen gebruik van
de canule(s) kan de prestaties negatief beïnvloeden.

Opmerking: Bestemd voor de leverancier van deapparatuur: De volgende accessoires voor toediening
van zuurstof worden aanbevolen voor gebruik met de
FreeStyle Comfort:• Neuscanule met 2,1 m (7 ft.) lange slang (max. aantall/min.: 6 l/min.): CAIRE-onderdeelnummer 5408-SEQ.
• Informatie over aanvullende aanbevolen accessoires
is online beschikbaar op www.caireinc.com.
5. Schakel de FreeStyle Comfort in door op de aan/uit-knop te drukken en selecteer uw voorgeschrevendebietinstelling met behulp van de debietknoppen
voor verhogen (+) of verlagen (-).

PN MN232-C4 L | Gebruikershandleiding NL - 217

FreeStyle Comfort
Ademhalingsfrequentie
15
20
25
30
35
40
Er klinkt een kort hoorbaar alarm om aan te geven
dat het apparaat is ingeschakeld.
Opmerking: Als de FreeStyle Comfort voor het eerst
wordt ingeschakeld, is de standaard debietinstelling 2.
Als de instelling wordt gewijzigd, wordt het apparaat
ingeschakeld met de vorige debietinstelling.
6. Plaats de neuscanule op uw gezicht inovereenstemming met de bij de canule geleverde
instructies van de fabrikant en adem normaal.
Telkens als u inademt, dient u via de neuscanulezuurstof toegediend te krijgen.
7. Om de debietinstelling te wijzigen, drukt u op dedebietknoppen voor verhogen (+) of verlagen (-)
tot het gewenste debiet. Het is normaal dat u een
ander geluid hoort wanneer u de debietinstellingenaanpast.

Instelling pulsvolume zonder autoSAT (ml)*
1
2
3
4
5
14,0
10,5
8,4
7,0
6,0
5,3

28,0
21,0
16,8
14,0
12,0
10,5

42,0
31,5
25,2
21,0
18,0
15,8

56,0
42,0
33,6
28,0
24,0
21,0

70,0
52,5
42,0
35,0
30,0
26,3

8. Schakel de FreeStyle Comfort uit door op de aan/uit-knop te drukken.
Opmerking: Het weergavescherm gaat naar de stand-bymodus en wordt na 20 seconden gedimd. Druk op een
willekeurige knop om het scherm uit de stand-by-modus
te halen.
9. Raadpleeg voor geavanceerde functies het menu
Navigatie-instructies.

BIJZONDERE PRODUCTEIGENSCHAPPEN
autoSAT®
De eigen autoSAT-functie van CAIRE stelt de
FreeStyle Comfort in staat om een consistentzuurstofvolume (bolus) per ademhaling toe te dienen,
tot een maximale ademhalingsfrequentie. Naarmate
de ademhalingsfrequentie van de gebruiker toeneemt
tijdens activiteiten zoals lopen, past de autoSAT-functie
de werking van het apparaat aan om een ingesteld
bolusvolume te handhaven.
UltraSense®
Deze gevoelige pulsdosistrigger detecteert dedrukverandering tijdens het ademhalen en geeft dezuurstofpuls af wanneer u inademt.

Instelling puslvolume met autoSAT (ml)*2345Ademha- 1lingsfrequentie
15
10,5
21,0
31,5
42,0
52,5
20
10,5
21,0
31,5
42,0
52,5
25
10,5
21,0
31,5
42,0
42,0
30
10,5
21,0
31,5
35,0
35,0
35
10,5
21,0
30,0
30,0
30,0
40
10,5
21,0
26,3
26,3
26,3
* De tolerantie is +/- 15%

218 - NL PN MN232-C4 L | Gebruikershandleiding

autoDOSE®
Als er een ‘geen adem’-alarm optreedt, geeft dezefunctie zuurstofpulsen af met 20 ademhalingen per
minuut bij de huidige debietinstelling van het apparaat
totdat de alarmtoestand is verholpen.

FreeStyle Comfort
Voedingen
De FreeStyle Comfort kan op drie manieren worden
aangedreven:• Herlaadbare accu
• Wisselstroom• Gelijkstroom
In de volgende rubrieken vindt u informatie overde werking van al deze systemen.
WAARSCHUWING: GEBRUIKSLECHTS DE VOEDINGEN DIE IN DEZEGEBRUIKERSHANDLEIDING WORDENOMSCHREVEN. HET GEBRUIK VAN
VOEDINGEN EN ACCU'S DIE NIET DOORCAIRE WORDEN VERSTREKT, DREIGTHET APPARAAT TE BESCHADIGENEN DE WERKING ERVAN TE HINDEREN.

DE ACCU VERWIJDEREN

Trek de accu-ontgrendelingsknop weg van de
FreeStyle Comfort. HOUD deze knop INGEDRUKTen schuif de accu intussen uit de concentrator door
hem ervan weg te trekken.
WAARSCHUWING: DE ZUURSTOFCONCENTRATOR MOET UIT STAAN EN ZIJNLOSGEKOPPELD VAN DE EXTERNE VOEDING VOORDAT U DE ACCU VERWIJDERT.

ACCU

De FreeStyle Comfort kan worden aangedreven doorde herlaadbare accu die bij het apparaat wordt
geleverd. Er gelden geen gebruiksbeperkingen voor
de concentrator wanneer deze op de accu werkt. Alle
debieten zijn beschikbaar bij aandrijving via de accu.

DE ACCULADING CONTROLEREN

DE ACCU PLAATSEN

Lijn de accu uit met de onderste behuizing vanuit het
midden van de FreeStyle Comfort. Duw de accu naarbinnen totdat u een klik hoort om de vergrendeling aan
te geven.

Om de acculading te controleren wanneer deze
in de FreeStyle Comfort is geplaatst, raadpleegt
u het accupictogram en de geschatte resterende tijd
op de weergave.
80–100%60–80%

40–60%

20–40%

20%

PN MN232-C4 L | Gebruikershandleiding NL - 219

FreeStyle Comfort
Opmerking: Als de voeding is aangesloten, wordt een
stekkerindicator weergegeven in plaats van het resterende
laadpercentage.
Om de acculading te controleren wanneer deze
niet in de FreeStyle Comfort is geplaatst, legt u deaccu op zijn rug en drukt u op de knop ‘TEST’.
De controlelampjes voor de acculading (25–100%) aan
de rechterkant van de knop ‘TEST’ gaan branden om
de acculading aan te geven.

ACCUDUUR IN WERKING

Wanneer de externe wisselstroom- ofgelijkstroomvoeding wordt losgekoppeld, schakelt deconcentrator automatisch over op het accuvermogen als
de accu is geplaatst en is opgeladen. In de onderstaande
tabel wordt de gebruikelijke werkingsduur van een
nieuwe accu getoond. De accuduur wordt niet beïnvloeddoor uw ademhalingsfrequentie.
Pulsinstelling

Duur van enkele(8-cellige) accu

Duur van dubbele(16-cellige) accu

1

tot 8 uur

tot 16 uur2

tot 4 uur

tot 8 uur3

tot 3 uur

tot 6 uur4

tot 2,25 uur

tot 4,5 uur

5

tot 2 uur

tot 4 uurGEBRUIKELIJKE OPLAADTIJD ACCU

De gebruikelijke tijd om uw accu volledig op te laden
vanaf een volledig lege accu is 3,5 uur voor een enkele
accu en 6,0 uur voor een dubbele accu.

EERSTE KEER OPLADEN

De nieuwe accu die bij uw FreeStyle Comfort wordt
geleverd, is slechts gedeeltelijk opgeladen wanneer
deze uit de fabriek komt. Vóór het eerste gebruik
moet u de accu volledig opladen met wisselstroomof gelijkstroom.

ALGEMENE LAADINFORMATIE

De FreeStyle Comfort zal de accu altijd opladen
wanneer deze in de concentrator is geplaatst
en wanneer de concentrator op een externe
wisselstroom- of gelijkstroomvoeding
is aangesloten (behalve in een vliegtuig).
De accu is volledig opgeladen wanneer 100% wordt
weergegeven.
Ook kan de accu met het optionele laadstation worden
opgeladen.

220 - NL PN MN232-C4 L | Gebruikershandleiding

Opmerking: De accu begint mogelijk niet meteen
met opladen als de accu volledig leeg is.

LEVENSDUUR ACCUDe FreeStyle Comfort-accu's zijn ontworpen
om 300 laad-/ontlaadcycli mee te gaan.

FreeStyle Comfort
Wisselstroom- en
gelijkstroomvoeding
De FreeStyle Comfort wordt standaard geleverd met
een wisselstroomvoeding en een gelijkstroomvoeding.
De wisselstroomvoeding omvat de volgende zaken:
	a. Wisselstroomvoedingsblok dat werkt tussen
100–240 V, 50–60 Hz.
	b. Wisselstroomsnoer om aan te sluiten
op het voedingsblok en de bijbehorende
wisselstroomuitgang.

De gelijkstroomvoeding bestaat uit een snoer dat
rechtstreeks op het apparaat en de gelijkstroombron
kan worden aangesloten.

2. Sluit het wisselstroomsnoer aan op het
wisselstroomvoedingsblok.

WISSELSTROOM

Wisselstroom is bedoeld voor gebruik in huis of waar
er standaard wisselstroom beschikbaar is. De FreeStyle
Comfort functioneert volledig op wisselstroom.
De FreeStyle Comfort laadt de accu ook op als er
wisselstroom beschikbaar is (behalve in een vliegtuig).
De accu wordt opgeladen, ongeacht of de FreeStyle
Comfort is in- of uitgeschakeld.
WAARSCHUWING: SLUIT GEEN EXTERNEVOEDING AAN ZONDER DAT ER EEN
ACCU IN DE ZUURSTOFCONCENTRATORIS GEPLAATST.
De FreeStyle Comfort aansluiten op wisselstroom:
1. Sluit het wisselstroomvoedingsblok aan op de
ingang van de stroomaansluiting aan de voorkant,
rechts onderaan de FreeStyle Comfort. Duw devoeding stevig in de aansluiting zodat hij goedvastzit.

3. Steek het andere uiteinde van het wisselstroomsnoer
in de muur of in een ander geschikt stopcontact.

PN MN232-C4 L | Gebruikershandleiding NL - 221

FreeStyle Comfort
4. Schakel de FreeStyle Comfort in door op de aan/uit-knop te drukken en selecteer uw voorgeschrevendebietinstelling met behulp van de debietknoppen
voor verhogen (+) of verlagen (-).

Opmerking: Als de accu moet worden opgeladen, heeft
het accupictogram op de weergave één oranje balk.

GELIJKSTROOM

Gelijkstroom is bedoeld voor gebruik op
voedingsaansluitingen van auto's, boten of andere
motorvoertuigen. De FreeStyle Comfort is volledig
functioneel voor gebruik op gelijkstroom. Ook laadt
de FreeStyle Comfort de accu zo nodig op zodra er
gelijkstroom beschikbaar is. De accu wordt opgeladen,
ongeacht of het apparaat is in- of uitgeschakeld of aldan niet wordt gebruikt.
WAARSCHUWING: SLUIT GEEN EXTERNEVOEDING AAN ZONDER DAT ER EEN
ACCU IN DE ZUURSTOFCONCENTRATORIS GEPLAATST.
1. Schakel uw motorvoertuig in.
2. Sluit de gelijkstroomvoeding aan op de voorkant
van de voedingspoort, rechts onderaan de FreeStyle
Comfort. Duw de voeding stevig in de aansluiting
zodat hij goed vastzit.

3. Steek het andere uiteinde van het gelijkstroomsnoer
in de voedingsaansluiting van de auto of in een
andere gelijkstroomaansluiting.

222 - NL PN MN232-C4 L | Gebruikershandleiding

4. Schakel de FreeStyle Comfort in door op de aan/uit-knop te drukken en selecteer uw voorgeschrevendebietinstelling met behulp van de debietknoppen
voor verhogen (+) of verlagen (-)
LET OP: Om mogelijke brand of
oververhitting te voorkomen, moet u ervoor
zorgen dat het 12VDC-stopcontact vrij isvan vuil en dat de adapterstekker van degelijkstroomvoeding goed past en stevig
op zijn plaats blijft.
LET OP: Metalen oppervlakken van gelijkstroomvoedingkunnen ZEER HEET worden wanneer ze in gebruik zijn
door hoge stroom. Laat de oppervlakken voldoende
afkoelen voordat u ze aanraakt.
LET OP: Gebruik de gelijkstroomvoeding niet met
extensies, splitters of adapters. Het gebruik van
niet-goedgekeurde accessoires kan leiden tot
oververhitting of beschadiging.
LET OP: Door het apparaat op een 12VDC-stopcontact
aan te sluiten, kan de batterij van de auto leegraken als de motor niet draait, zelfs als het apparaat
uitgeschakeld is.
LET OP: Start een voertuig niet met startkabels terwijl
de gelijkstroomvoeding is aangesloten.

FreeStyle Comfort
Reiniging, zorg en routineonderhoud
ROUTINEONDERHOUDPreventief onderhoud
Het apparaat is onderhoudsvrij. Filters in het
hulpmiddel zijn bedoeld om de levensduur van het
hulpmiddel te verlengen. Als het apparaat niet goedwerkt, moet het apparaat worden teruggestuurd naarde leverancier voor reparatie. Onderhoud aan interne
onderdelen binnen in de behuizing van de FreeStyle
Comfort moet door de zuurstofleverancier worden
uitgevoerd.
De gebruiker van het apparaat is verantwoordelijk voor
de onderstaande onderhoudsprocedure.
• Wekelijks: reiniging van de luchtinlaatfilters
WAARSCHUWING: PROBEER DE PLASTIC
DEKSELS VAN DE ZUURSTOFCONCENTRATOR NIET TE OPENEN OF TE VERWIJDEREN. HET APPARAAT MAG SLECHTSDOOR EEN BEVOEGDE ONDERHOUDSTECHNICUS WORDEN GEOPEND.

LUCHTINLAATFILTERS

Lucht komt de FreeStyle Comfort binnen via
luchtinlaatfilters aan de voorkant, onderkant van dezuurstofconcentrator. Deze filters voorkomen dat stof
en andere grote deeltjes uit de lucht in het apparaat
terechtkomen.
De luchtinlaatfilters moeten minstens eenmaal per
week worden gereinigd. De leverancier van uw
zuurstofapparatuur raadt u mogelijk aan het filter vaker
te reinigen, naargelang de gebruiksomstandigheden.
Volg deze stappen om de luchtinlaatfilters correct
te reinigen:
1. Verwijder de twee filters aan de voorkant van
de FreeStyle Comfort door het bijbehorende lipjeomhoog te duwen en naar buiten te trekken.

WAARSCHUWING: GEBRUIK DE ZUURSTOFCONCENTRATOR NIET ALS HETLUCHTINLAATFILTER NIET IS BEVESTIGD.

ONDERHOUD ACCU

Stel de accu niet bloot aan vloeistoffen.
Probeer de accu niet te demonteren of te openen.
Stel de accu niet bloot aan open vlammen.
Laat de accu niet vallen.
Houd de accu buiten het bereik van kinderen.
Als uw accu beschadigd lijkt, neem dan contact op met
de leverancier van uw zuurstofapparatuur vóór gebruik.
Plaats de accu niet in of in de buurt van een magnetronof andere kookapparaten.

OPSLAG ACCU

De accu van de zuurstofconcentrator hoeft niet
volledig leeg te zijn alvorens deze weer op te laden.
Het is aanbevolen de accu van de zuurstofconcentrator
na elk gebruik op te laden.
Stel de accu niet bloot aan overmatige hitte, zoals in
rechtstreeks zonlicht of in een auto. Hierdoor dreigen
de prestaties en levensduur van de accu erop achteruit
te gaan. Ook dreigt de accu hierdoor hitte af te geven,te roken, vuur te vatten of te ontploffen. De accu moetin een stabiele omgeving worden bewaard, met eenlage vochtigheidsgraad (relatieve vochtigheid vanminder dan 70%) en zonder corrosieve gassen. Om
roest te voorkomen, moet u omstandigheden waarincondensvorming kan optreden vermijden, zoals snelle
schommelingen van de omgevingstemperatuur.
Voor opslag gedurende een maand of langer wordt een
ontladen accu of een deels geladen accu tussen 50% en
75% (twee tot drie leds) aanbevolen. Haal de accu uit
de zuurstofconcentrator zodra deze is opgeladen.

2. Was beide filters in een warme oplossing van zeep
en water.
3. Spoel de filters grondig en verwijder overtollig
water met een zachte, stofvrije doek. Controleer
of de filters volledig droog zijn alvorens ze terug
te plaatsen.
4. Plaats de schone, droge filters terug.

PN MN232-C4 L | Gebruikershandleiding NL - 223

FreeStyle Comfort
REINIGING

Schakel het apparaat UIT en koppel het los van wisselof gelijkstroomvoeding voordat u het schoonmaakt.
Spuit NIET rechtstreeks op de behuizing. Gebruik een
vochtige (geen natte) doek of spons. Bespuit de doekof spons met een mild huishoudelijk reinigingsmiddel
om de behuizing en voedingsbron schoon te maken.
Ga te werk overeenkomstig de aanbevelingen vande fabrikant van het schoonmaakmiddel.
WAARSCHUWING: SCHAKEL HETAPPARAAT UIT EN TREK HET SNOERUIT HET STOPCONTACT ALVORENSHET APPARAAT TE REINIGEN OM
ONOPZETTELIJKE ELEKTRISCHESCHOKKEN EN BRANDGEVAAR TEVOORKOMEN. ALLEEN DE LEVERANCIER
VAN UW APPARATUUR OF EEN
BEVOEGDE SERVICEMONTEUR MOGEN
DE DEKSELS VERWIJDEREN OF HETAPPARAAT ONDERHOUDEN.
WAARSCHUWING: U MOET EROVER WAKEN DAT DEZUURSTOFCONCENTRATOR EN DE ACCU NIET NAT
WORDEN EN DAT ER GEEN VLOEISTOFFEN IN HETAPPARAAT TERECHTKOMEN. HIERDOOR DREIGTHET APPARAAT EEN STORING TE VERTONEN OF
UIT TE VALLEN. BOVENDIEN NEEMT HET RISICO OP
ELEKTRISCHE SCHOKKEN OF BRANDWONDEN TOE.
WAARSCHUWING: GEBRUIK GEEN VLOEISTOF
RECHTSTREEKS OP HET APPARAAT. HIERONDER
VINDT U EEN NIET-ALLESOMVATTENDE LIJSTVAN ONGEWENSTE CHEMISCHE STOFFEN:
ALCOHOL EN ALCOHOLHOUDENDE PRODUCTEN,GECONCENTREERDE CHLOORHOUDENDE
PRODUCTEN (ETHYLEENCHLORIDE) EN
OLIEHOUDENDE PRODUCTEN. DEZE MOGEN NIETWORDEN GEBRUIKT OM DE PLASTIC BEHUIZINGVAN DE CONCENTRATOR TE REINIGEN, AANGEZIENDEZE HET APPARAAT KUNNEN BESCHADIGEN.
WAARSCHUWING: REINIG DE BEHUIZING, HET
BEDIENINGSPANEEL EN HET NETSNOER UITSLUITEND MET EEN MILDE HUISHOUDREINIGER DIE UAANBRENGT MET EEN VOCHTIGE (GEEN NATTE)DOEK OF SPONS. DROOG NADIEN HET VOLLEDIGEOPPERVLAK. ZORG DAT ER GEEN VLOEISTOF IN
DE CONCENTRATOR KOMT. BESTEED BIJZONDEREAANDACHT AAN DE ZUURSTOFUITGANG VOOR DE
CANULEAANSLUITING EN ZORG ERVOOR DAT DEZE
ALTIJD VRIJ IS VAN STOF, WATER EN DEELTJES.

224 - NL PN MN232-C4 L | Gebruikershandleiding

WAARSCHUWING: GEBRUIK EEN VOCHTIGE (GEEN
NATTE) DOEK OF SPONS OM DE ACCU TE REINIGEN.
SPUIT ALLEREERST EEN MILDE HUISHOUDREINIGEROP DE DOEK OF SPONS EN REINIG DE
ACCUBEHUIZING.
WAARSCHUWING: NIET REINIGEN MET EENGEAUTOMATISEERD WASSYSTEEM.

DE CANULE VERVANGEN

Volg altijd de instructies voor correct gebruik van
de fabrikant van de canule. Vervang de neuscanuleof zuurstofslang elke 3 tot 6 maanden (bij dagelijks
gebruik, 3 maanden) volgens de aanbevelingen vande fabrikant, zoals aanbevolen door de fabrikant vande canule of de leverancier van uw zuurstofapparatuur.
Uw arts of zuurstofleverancier zal u informatie gevenover reiniging en vervanging.
Extra vervangonderdelen zijn verkrijgbaar bij
de leverancier van uw zuurstofapparatuur.

FreeStyle Comfort
Alarmsituaties
Prioriteit nr.

Alarmbeschrijving

1

Systeemstoring hoge
batterijtemperatuur

Weergave

HoorbaarsignaalMeer details/aanbevolen handeling

1 pieptoon elke
20 sec.

Het apparaat slaat alarm als de accutemperatuur de veiligheidsuitschakelgrens nadert.
Als de uitschakelgrens is bereikt, wordt
het apparaat uitgeschakeld. Als de externe
voeding is aangesloten, verschijnt het alarmscherm en is het hoorbare alarm gedurende
2 minuten te horen.
Test met een accu op kamertemperatuur.
Neem contact op met de leverancier als het
probleem blijft aanhouden.

2

Systeemstoring hoge
temperatuur

1 pieptoon elke
20 sec.

Het apparaat slaat alarm als de interne
temperatuur de uitschakelgrens nadert.
Als de uitschakelgrens is bereikt, wordt
het apparaat uitgeschakeld en verschijnt
het alarmscherm en is het hoorbare alarm
gedurende 2 minuten te horen.
Controleer of er geen openingen geblokkeerd
zijn. Neem contact op met de leverancier als
het probleem blijft aanhouden.

3

Systeemstoring hoge
druk (hogesysteemdrukof compressorstoring)

1 pieptoon elke
20 sec.

4

Systeemstoring debiet(OCD-klepstoring, obstructie
van gasbaan)

1 pieptoon elke
30 sec.

Neem contact op met de leverancier als het
probleem blijft aanhouden.

5

Uitschakelingdoor bijna lege
batterij

1 pieptoon elke
30 sec.

Het alarm is hoorbaar totdat de accu leegis en het apparaat wordt uitgeschakeld.

Als de uitschakelgrens is bereikt, wordt
het apparaat uitgeschakeld en verschijnt
het alarmscherm en is het hoorbare alarm
gedurende 2 minuten te horen.
Neem contact op met de leverancier als het
probleem blijft aanhouden.

Laad de accu op.

PN MN232-C4 L | Gebruikershandleiding NL - 225

FreeStyle Comfort
Prioriteit nr.

Alarmbeschrijving

6

Geen
ademhaling
waargenomen

Weergave

HoorbaarsignaalMeer details

1 pieptoon elke
10 minuten

Geen ademhaling gedetecteerd gedurende
15 seconden.
Geactiveerd na de eerste ademhaling of het
opwarmscherm wordt uitgeschakeld, afhankelijk van wat eerder komt.
Controleer of de canule knikken of lekken
vertoont. Verhoog de gevoeligheidsinstelling.
Neem contact op met de leverancier als het
probleem blijft aanhouden.

7

OpwarmenGeen

Het apparaat begint met automatisch doseren
totdat een ademhaling wordt gedetecteerdof het opwarmen eindigt, afhankelijk van wat
eerder komt.

8

Lage
concentratie

1 pieptoon elke
10 minuten, tot
30 min.
lang

Neem contact op met de leverancier als het
probleem blijft aanhouden.

9

Waarschuwing
batterij bijnaleeg

1 pieptoon,geen
herhaling

Laad de accu op.

10

Ademhalingsfrequentie
overschreden

1 pieptoon,geen
herhaling

De ademhalingsfrequentie overschrijdt
40 bpm.

11

Systeemstoring ventilatorstoring

Geen

Neem contact op met de leverancier als het
probleem blijft aanhouden.

12

Systeemstoring lage druk

Geen

Neem contact op met de leverancier als het
probleem blijft aanhouden.

13

Systeemstoring lage
temp.

Geen

Laat het apparaat opwarmen tot op
kamertemperatuur. Neem contact op met deleverancier als het probleem blijft aanhouden.

226 - NL PN MN232-C4 L | Gebruikershandleiding

Als de dosis ten onrechte wordt getriggerd,
stel de gevoeligheidsinstelling dan lager in.
Neem contact op met de leverancier als het
probleem blijft aanhouden.

FreeStyle Comfort
Prioriteit nr.

Alarmbeschrijving

Info

Opstarten

Weergave

HoorbaarsignaalMeer details

1 infopieptoon

Het scherm verschijnt bij het inschakelen
en u hoort één pieptoon om te bevestigen
dat de zoemer functioneert.
Het startscherm toont de huidige
firmwarerevisie in de rechteronderhoek.

InfoNormaal bedrijfsscherm in
autoSAT-modus (geengelijkstroom
aangesloten)

Geen

Info

Normaalbedrijfsscherm
met autoSAT
gedeactiveerd (geengelijkstroom
aangesloten)

Geen

Info

Instellinggewijzigd

1 infopieptoon

Info

Sluit de externe
voeding aan inde uit-stand

Geen

Info

Sluit deexternevoedingaan in de
bedrijfsstand

Geen

Info

Koppel deexternevoeding los inde uit-stand

1 infopieptoon

Het CAIRE-scherm verschijnt gedurende
2 sec., gevolgd door de accu- en stekkerstatus gedurende 5 sec. voordat het schermwordt uitgeschakeld.

Laat het CAIRE-scherm 2 seconden
verschijnen en schakel daarna het scherm uit.

PN MN232-C4 L | Gebruikershandleiding NL - 227

FreeStyle Comfort
Prioriteit nr.

Alarmbeschrijving

Weergave

Info

Koppel deexterne voeding los in debedrijfsstand(met accu
geplaatst)

1 infopieptoon

Info

In de standby-modus:
wanneer op
de omhoog- of
omlaag-knop
wordt gedrukt,
toont hetscherm deaccu- enlaadstatus

Geen

Info

Vliegtuigmodus op
accuvoeding

1 langepieptoon

Info

Vliegtuigmodus bij aansluiting op externe
voeding1 langepieptoon

228 - NL PN MN232-C4 L | Gebruikershandleiding

HoorbaarsignaalMeer details

Het scherm verschijnt gedurende 5 sec.
en wordt vervolgens uitgeschakeld.

FreeStyle Comfort
Menu Navigatie-instructies
U kunt het menu raadplegen in de stand-by-modus
of de bedrijfsmodus. Als er in de bedrijfsmodus
een alarm optreedt, wordt het menu automatischafgesloten. Als er 20 seconden geen knop wordt
ingedrukt, wordt het menu automatisch afgesloten.
In stand-by-modus moet u mogelijk tweemaal demenuknop indrukken om het apparaat uit de slaapstandte halen, waarna u het menu kunt raadplegen.
Er zijn vier menuschermen:
1. Statusscherm: Hier worden het
firmwarerevisienummer, de bedrijfstijd
en de accu-ontladingscycli weergegeven

Vliegtuigmodus

2. Vliegtuigmodus (AAN/UIT)
3. Gevoeligheid: instelbaar (1-5)
4. autoSAT (AAN/UIT)

Gevoeligheid

Statusscherm ter illustratie

BEDIENING INSTELLINGENMENU
U kunt het Instellingenmenu raadplegen door in debedrijfs- of stand-by-modus op de knop Menu/Selecteren

autoSATte drukken.
Druk opom door de menu's te scrollen.

Statuspagina

PN MN232-C4 L | Gebruikershandleiding NL - 229

FreeStyle Comfort
STATUSPAGINA
De statuspagina van de FreeStyle Comfort toont defirmwarerevisie, bedrijfsuren en het aantal batterijcycli
op de aangesloten batterij.

Vliegtuigmodus aan bevestigd

Om de vliegtuigmodus uit te schakelen, herhaaltu de bovenstaande stappen terwijl u ‘UIT’ selecteert.
Statuspagina

Bedrijfsuren worden weergegeven in hele uren.
Batterijontladingen zijn het aantal cycli dat deaangesloten batterij is leeggelopen en opnieuw isopgeladen.

Als u opdrukt wanneer op het scherm het
symbool ‘Volgende’ verschijnt, schakelt demenupagina naar de volgende schermoptie.

VLIEGTUIGMODUS

De vliegtuigmodus is bedoeld om te worden gebruikt
in de bedrijfsmodus. De vliegtuigmodus wordt
automatisch uitgeschakeld wanneer het apparaat
stand-by blijft op de accuvoeding.

Om de FreeStyle Comfort te vervoeren, wordt devliegtuigmodus uitgeschakeld.

Opmerking: Raadpleeg de rubriek ‘De vliegtuigmodus
activeren’ voor meer informatie over de vliegtuigmodus.

GEVOELIGHEIDSMODUS
De FreeStyle Comfort wordt vervoerd met een standaardfabrieksinstelling van 2 voor de gevoeligheidsmodus en
dit hoeft niet te worden aangepast in normale omstandigheden. De gevoeligheidsinstelling kan worden ingesteld
van 1 tot 5, waarbij 1 het minst gevoelig en 5 het meest
gevoelig is.
Hoofdscherm vliegtuigmodus

Om de modus te activeren, drukt u op
terwijl het
pictogram WIJZIGEN zichtbaar is. Het blauwe woord‘AAN’ verschijnt.

Gevoeligheidsmodusscherm bij instelling 2

Om de gevoeligheidsmodus aan te passen, drukt u op
terwijl het pictogram WIJZIGEN zichtbaar is.
De waarde verspringt naar het volgende blauwe nummer.
Vliegtuigmodus aan geselecteerd

Om vliegtuigmodus aan te bevestigen, drukt u op
terwijl het pictogram OPSLAAN zichtbaar is. Het
zwarte woord ‘AAN’ verschijnt.
Om de vliegtuigmodus te annuleren, drukt u op
terwijl het pictogram ANNULEREN zichtbaar is.

230 - NL PN MN232-C4 L | Gebruikershandleiding

FreeStyle Comfort
AUTOSAT-MODUS
De FreeStyle Comfort wordt vervoerd met
de autoSAT-modus ingeschakeld.

Gevoeligheidsmodusinstelling 3 geselecteerd

Om de instelling 3 in de gevoeligheidsmodus tebevestigen, drukt u op
terwijl het pictogram
OPSLAAN zichtbaar is. De waarde ‘3’ wordt in het
zwart weergegeven.
Om de wijziging van de gevoeligheidsmodus
te annuleren, drukt u op
terwijl het pictogram
ANNULEREN zichtbaar is.

Hoofdscherm autoSAT-modus

Om de modus uit te schakelen, drukt u op
terwijlhet pictogram WIJZIGEN zichtbaar is. Het blauwe
woord ‘UIT’ verschijnt.

autoSAT-modus Uit geselecteerd
Gevoeligheidsmodusinstelling 3 bevestigd

Om de uitschakeling van de autoSAT-modus tebevestigen, drukt u op
terwijl het pictogram
OPSLAAN zichtbaar is. Het zwarte woord ‘UIT’
verschijnt.
Om de inschakeling van de autoSAT-modus
te annuleren, drukt u op
terwijl het pictogram
ANNULEREN zichtbaar is.

Gevoeligheidsinstelling 5 (max.) bevestigd

Als u opdrukt wanneer op het scherm het
symbool ‘Volgende’ verschijnt, schakelt demenupagina naar de volgende schermoptie.

autoSAT-modus uit bevestigd

Om de autoSAT-modus in te schakelen, herhaaltu de bovenstaande stappen terwijl u ‘AAN’ selecteert.
Als u opdrukt wanneer op het scherm het
symbool ‘Volgende’ verschijnt, schakelt demenupagina naar de volgende schermoptie.

PN MN232-C4 L | Gebruikershandleiding NL - 231

FreeStyle Comfort
Probleemoplossing
Probleem

Mogelijke oorzaak

OplossingDe FreeStyle Comfort toont
een alarmbericht en laat een
niet-continue pieptoon horen.

Raadpleeg de tabel met
alarmsituaties.

Raadpleeg de tabel met alarmsituaties.

De zuurstofconcentrator kan
niet worden aangezet.

De accu is ontladen.

Sluit het apparaat aan op een gelijkstroomof wisselstroombron.

Niet verbonden met een
stroombron.
Algemene storing.

Controleer of de uitwendige aansluitingen goedzijn bevestigd.
Voer een volledige opstartcyclus uit door de accu
te verwijderen en een eventuele externe wissel- of
gelijkstroomvoedingsbron los te koppelen. Plaats deaccu vervolgens opnieuw, waarna de opstartcyclus
voltooid is. Sluit indien gewenst de externe
voedingsbron opnieuw aan.
Neem contact op met de leverancier van
uw apparaat en schakel zo nodig over op een
alternatieve zuurstofbron.

Alle overige problemen.

Voer een volledige opstartcyclus uit door de accu
te verwijderen en een eventuele externe wissel- of
gelijkstroomvoedingsbron los te koppelen. Plaats deaccu vervolgens opnieuw, waarna de opstartcyclus
voltooid is. Sluit indien gewenst de externe
voedingsbron opnieuw aan.
Schakel over op een alternatieve zuurstofbronen neem contact op met de leverancier van uw
apparatuur.

Opmerking: Bepaalde alarmen zorgen ervoor dat het apparaat wordt uitgeschakeld.
Opmerking: Tijdens een volledige stroomuitval wordt het alarmsysteem opnieuw ingesteld (eerdere alarmen worden daarbij
gewist) en verschijnt een alarm alleen als er nog steeds sprake is van de alarmsituatie zodra de voeding is hersteld.

232 - NL PN MN232-C4 L | Gebruikershandleiding

FreeStyle Comfort
Reisinformatie
De FreeStyle Comfort voldoet aan de FAA-richtlijnen
voor gebruik aan boord van commerciële vliegtuigen.
Vóór elke reis (vliegtuig, trein, cruiseschip, enz.) moet u:• De goede werking van de concentrator controleren.
• Minstens 48 uur voor vertrek contact opnemen met
de reismaatschappij om te laten weten dat u met een
FreeStyle Comfort zult reizen.
Het is aanbevolen om altijd extra accu's bij u te hebben
wanneer u reist. Bij vliegreizen moet de accu voldoende
opgeladen zijn om de tijd op de grond (inchecken,
veiligheidscontroles, enz.) en tussenlandingen te dekken.
De meeste luchtvaartmaatschappijen vereisen dat u overeen geschikt aantal volledig opgeladen accu's beschiktom het toestel van stroom te voorzien gedurende
minstens 150% van de verwachte maximale duur van dereis (vluchttijden, tijd op de grond voor en na de vlucht,
tijdens verbindingen en voor onverwachte vertragingen).

1. Schakel de POC in met behulp van ALLEEN
accuvoeding als er geen externe voedingsbron isaangesloten.

2. Schakel de vliegtuigmodus in met behulp van een
van de volgende methoden:
a. Volg de instructies om de vliegtuigmodus in teschakelen in het menu Navigatie-instructies.

Opmerking: Als u vliegt, is het raadzaam extra accu's
in een veilige draagtas mee te nemen.

FAA-REGELGEVING

Reserveaccu's moeten, om kortsluiting tevoorkomen, individueel worden beschermd door
ze in de oorspronkelijke verpakking te plaatsen of
de aansluitingen op een andere manier te isoleren,bijv. door blootliggende aansluitingen te bedekken
met tape of door elke accu in een aparte plastic zakof beschermhoes te doen, en mogen alleen in dehandbagage worden vervoerd.

b. Houd de omhoog- en omlaag-knop 5 seconden
ingedrukt totdat u een lange pieptoon hoort en het
vliegtuigsymbool ziet.

DE VLIEGTUIGMODUS ACTIVEREN
Accuvoeding
De POC kan in een vliegtuig ALLEEN met behulp vande accuvoeding worden gebruikt volgens de normale
bedieningsprocedures als er geen externe voedingsbron
is aangesloten.

3. Wanneer het vliegtuigsymbool verschijnt,
sluit u de POC aan.

Externe voeding
Als in een vliegtuig een elektrisch stopcontactbeschikbaar is dat werkt en de locatie daarvan geschikt
is voor gebruik als een POC-voedingsbron, kan dit
dienen als back-up voor POC-accu's. In dit geval moet
de volgende aanbevolen procedure met betrekking tot
de overgang van de accuvoeding naar de elektrischevoeding van het vliegtuig worden gevolgd:

Opmerking: Vanwege vliegtuigbeperkingen kunnende accu's niet worden opgeladen in de vliegtuigmodus.

PN MN232-C4 L | Gebruikershandleiding NL - 233

FreeStyle Comfort
Accessoires

HET APPARAAT IN DE DRAAGTASPLAATSEN

Met het oog op correcte prestaties en veiligheid mag
u alleen de opgesomde accessoires gebruiken die via
de leverancier van uw zuurstofapparatuur door CAIRE
worden verstrekt. Het gebruik van accessoires die niet
op de onderstaande lijst staan, kan de prestaties en/of
veiligheid van de concentrator negatief beïnvloeden.

1. Plaats het apparaat tot onderin de tas en stem defilter- en accuopeningen juist af.

Standaardaccessoires van de FreeStyle Comfort
Draagtas

MI396-1

RugzakMI459-1

Schouderriem

MI397-1

Wisselstroomvoeding(alle snoeren inbegrepen)

PW036-1S

Gelijkstroomsnoer

CD041-1

Externe groffilter (rechts)

FI226-1

Externe groffilter (links)

FI227-1

2. Doe de rits onderaan de tas dicht.

Optionele accessoires beschikbaar
Extra 8-cellige accu

BT034-1S

16-cellige accu

BT035-1S

Acculaadstation

BT036-1

Ruime accessoiretas

MI372-2

Opmerking: Er zijn mogelijk bijkomende opties
beschikbaar voor landspecifieke snoeren, daar waar dit
hierboven is aangegeven. Neem contact op met CAIRE
of met de leverancier van uw zuurstofapparatuur als u
alternatieve opties nodig hebt voor een bestelling.

Instructies draagtas
Volg de onderstaande instructies om uw FreeStyle
Comfort veilig in de draagtas te plaatsen.

234 - NL PN MN232-C4 L | Gebruikershandleiding

DE SCHOUDERRIEM BEVESTIGEN
1. De schouderriem kan direct aan het apparaat of aan
de draagtas worden bevestigd.
2. Maak de riem vast aan de ring op het apparaat
of de tas.

FreeStyle Comfort
Verwijdering en hergebruik
Lever de FreeStyle Comfort, inclusief alle onderdelen, altijd in bij de leverancier van uw apparaat voorde juiste afvoer ervan. U kunt ook contact opnemenmet de lokale autoriteiten voor inlichtingen over decorrecte afvoer van de batterij. Het apparaat kan worden hergebruikt tussen personen, maar moet worden
teruggestuurd naar de leverancier van het apparaat
voor herverwerking.

AEEA en RoHS
Dit symbool herinnert gebruikers eraan om
de apparatuur aan het einde van de levensduur naar
een recyclingbedrijf te brengen, conform de AEEArichtlijn (afgedankte elektrische en elektronische
apparatuur).

Materialen in rechtstreeks contact
met de gebruiker• Behuizing concentrator: PC/ABS• Bedieningspaneel concentrator: Polyester EBG7
of gelijkwaardig• Zuurstofuitlaat, mondstuk: Aluminium• Etiket apparaat: Lexaan• Netsnoeraansluitingen: Polycarbonaat/vinylchloride• Netsnoer: PVC, metaal• Voeding: Lexaan 940 (polycarbonaat)• Accu: PC/ABS• Etiketten accu, voeding: Polyesterfolie• Draagkoffer concentrator: Cordura, neopreen, nylon
voering• Riem: Polypropyleen, acetyl, vinyl, Cordura,
neopreen

Onze producten voldoen aan de beperkingen van derichtlijn inzake gevaarlijke stoffen (RoHS). Ze bevatten geen sporen van lood of andere gevaarlijke stoffen.
LET OP: Neem contact op met de leverancier
van uw apparatuur of met uw gemeente voor
afvalverwerkingsinstructies om het apparaat
op een correcte manier af te voeren.
LET OP: Voer de moleculaire zeef altijd af inovereenstemming met alle nationale, provinciale
en lokale milieuvoorschriften.

PN MN232-C4 L | Gebruikershandleiding NL - 235

FreeStyle Comfort
EMC-testen
Leidraad en Verklaring van de fabrikant – Elektromagnetische emissies
De FreeStyle Comfort is bedoeld voor gebruik in de elektromagnetische omgeving die hieronder wordt gespecificeerd.
De klant of de gebruiker van de FreeStyle Comfort dient te verzekeren dat ze in een dergelijke omgeving worden gebruikt.
Emissietest
RF-emissies
CISPR 11
RF-emissies
CISPR 11
Harmonische emissies
IEC 61000-3-2
Spanningsfluctuaties/
flikkeremissies
IEC 61000-3-3

Naleving

Elektromagnetische omgeving – Leidraad

Groep 1De FreeStyle Comfort gebruikt alleen RF-energie voor zijninterne werking. De RF-emissies zijn daarom erg laag en het is
onwaarschijnlijk dat deze enige storing zouden veroorzaken inelektronische apparatuur die zich dichtbij bevindt.

Klasse B
Klasse A
Conform

236 - NL PN MN232-C4 L | Gebruikershandleiding

De FreeStyle Comfort is geschikt voor gebruik in alle instellingen,waaronder instellingen met een woonfunctie en instellingen die
rechtstreeks zijn aangesloten op een laagspanningsnetwerk dat
gebouwen met een woonfunctie van stroom voorziet.

FreeStyle Comfort
Leidraad en Verklaring van de fabrikant – Elektromagnetische immuniteit

De FreeStyle Comfort is bedoeld voor gebruik in de elektromagnetische omgeving die hieronder wordt gespecificeerd.
De klant of de gebruiker van de FreeStyle Comfort dient te verzekeren dat ze in een dergelijke omgeving worden gebruikt.
Immuniteitstest
IEC 60601-testniveau
Nalevingsniveau
Elektromagnetische omgeving –Leidraad
Houten, betonnen of keramische tegelvloer
Elektromagnetische±8 kV contact±8 kV contact
vereist. Indien de vloerbedekking bestaat
omgeving – Leidraad
uit synthetisch materiaal, dient de relatieve±15 kV luchtafvoer±15 kV luchtafvoer
IEC 61000-4-2
luchtvochtigheid minimaal 30% te zijn.
±2 kV voor
±2 kV voor
voedingskabels
Snelle elektrische
De kwaliteit van de netspanning moet
voedingskabelstransiënten/lawines
die zijn van een typische professionele
zorginstelling en thuiszorgomgevingen.
IEC 61000-4-4±1 kV voor ingangs-/
N.v.t.
uitgangsleidingen±1 kV lijn(en) naar lijn(en)Stroomstoten
IEC 61000-4-5

Spanningsdalingen,korte
onderbrekingen en
spanningsfluctuaties
bij ingangslijnen van
de stroomvoorziening
IEC 61000-4-11±2 kV gemeenschappelijke ±1 kV lijn(en) naar
modus oplijn(en)wisselstroomlijnen

De kwaliteit van de netspanning moet
die zijn van een typische professionele
zorginstelling en thuiszorgomgevingen.

±1 kV differentieel op
wisselstroomlijnen±2 kV gemeenschappelijke
modus op I/O-buitenlijnen
0% UT gedurende0,5 cyclus (0°, 45°, 90°,135°, 180°, 225°, 270°,315°)

0% UT gedurende0,5 cyclus (0°, 45°, 90°,135°, 180°, 225°, 270°,315°)

0% UT gedurende
250/300 cycli (0°)

0% UT gedurende
250/300 cycli (0°)

De kwaliteit van de netspanning moet
die zijn van een typische professionele
zorginstelling en thuiszorgomgevingen.
0% UT gedurende
0% UT gedurendeIndien de gebruiker van de FreeStyle
1 cyclus (0°)
1 cyclus (0°)
Comfort een constante werking vereist
gedurende stroomonderbrekingen, wordt
70% UT (30% daling in UT) 70% UT (30% daling in UT) het aanbevolen de FreeStyle Comfort aan
gedurende 25/30 cycli (0°) gedurende 25/30 cycli (0°) te sluiten op een niet-onderbreekbare
stroomtoevoer of een batterij.

Netfrequentie
De magnetische velden van denetfrequentie dienen die van een(50/60 Hz)
3 A/m
3 A/m
gangbare professionele zorginstellingmagnetisch veld
IEC 61000-4-8
en thuiszorgomgevingen te zijn.
OPMERKING: UT is de spanning van de wisselstroomnetvoeding voorafgaand aan de toepassing van het testniveau.

OPMERKING: In sommige gevallen is het mogelijk dat de FreeStyle Comfort opnieuw moet worden ingesteld na een
aanzienlijke elektrostatische ontlading en kan het apparaat een volledige opstartcyclus vereisen om de normale werking
te kunnen hervatten. Zie het deel Probleemoplossing voor meer informatie.

PN MN232-C4 L | Gebruikershandleiding NL - 237

FreeStyle Comfort
Leidraad en Verklaring van de fabrikant – Elektromagnetische immuniteit

De FreeStyle Comfort is bedoeld voor gebruik in de elektromagnetische omgeving die hieronder wordt gespecificeerd.
De klant of de gebruiker van de FreeStyle Comfort dient te verzekeren dat ze in een dergelijke omgeving worden gebruikt.
Elektromagnetische omgeving –Immuniteitstest
IEC 60601-testniveau
Nalevingsniveau
Leidraad
10 Vrms
10 Vrms
Draagbare en mobiele RF-communicatieGeleide RF
apparatuur mag niet dichter bij onderdelen
IEC 61000-4-6
150 kHz tot 80 MHz
van de FreeStyle Comfort, inclusief dekabels, worden gebruikt dan de aanbevolen
scheidingsafstand die werd berekend aan
de hand van de vergelijking die toepasselijk
Uitgestraalde RF 10 V/m
10 V/mis op de frequentie van de zender.
IEC 61000-4-3
80 MHz tot 2700 MHz
Aanbevolen scheidingsafstandd = 1.2 P

d = 1.2 P = 80 MHz tot 800 MHz
d = 1.2 P = 800 MHz tot 2700 MHz
waarbij P staat voor het nominale maximale
uitgangsvermogen van de zender inwatt (W) volgens de fabrikant van de zender
en d voor de aanbevolen scheidingsafstandin meter (m).
Veldsterktes van stationaire RF-zenders,zoals bepaald door een overzicht van het
elektromagnetisch veld,a dienen lager
te zijn dan het nalevingsniveau in elk
frequentiebereik. b
Interferentie kan voorkomen in de nabijheid
van apparatuur waarop het volgende
symbool staat:

OPMERKING 1: Bij 80 MHz en 800 MHz geldt het hogere frequentiebereik.
OPMERKING 2: Het kan zijn dat deze richtlijnen niet op alle situaties van toepassing zijn. Elektromagnetische propagatie
wordt beïnvloed door absorptie en reflectie van gebouwen, voorwerpen en mensen.
aVeldsterktes van stationaire zenders zoals basisstations voor (mobiele/draadloze) radiotelefoons en landmobiele radio's,
amateurradio-, AM- en FM-radio-uitzendingen en televisie-uitzendingen, kunnen niet nauwkeurig theoretisch worden
voorspeld. Om de elektromagnetische omgeving van stationaire RF-zenders te evalueren, kan men een elektromagnetischterreinonderzoek uitvoeren. Wanneer de gemeten veldsterkte op de plek waar de FreeStyle Comfort gebruikt wordt boven
het hierboven weergegeven RF-nalevingsniveau uitkomt, moet de FreeStyle Comfort in de gaten worden gehouden om
te controleren of het apparaat normaal werkt. Als u constateert dat het apparaat niet normaal werkt, kan het nodig zijn om
extra maatregelen te treffen, zoals het anders richten of het verplaatsen van de FreeStyle Comfort.
bBinnen het frequentiebereik van 150 kHz tot 80 MHz dienen de veldsterktes minder dan 10 V/m te zijn.

238 - NL PN MN232-C4 L | Gebruikershandleiding

FreeStyle Comfort
Aanbevolen scheidingsafstand tussen draagbare en mobiele
RF-communicatieapparatuur en de FreeStyle Comfort

De FreeStyle Comfort is bedoeld voor gebruik in een elektromagnetische omgeving waarin uitgestraalde RF-storingen
worden beheerst. De klant of de gebruiker van de FreeStyle Comfort kan elektromagnetische storing helpen voorkomen
door een minimale scheidingsafstand tussen de draagbare en mobiele RF-communicatieapparatuur (zenders) en de
FreeStyle Comfort in acht te nemen zoals hieronder aanbevolen, overeenkomstig het maximale uitgangsvermogen vande communicatieapparatuur.
Nominaal maximaal
Scheidingsafstand volgens frequentie van de zenderuitgangsvermogen vanmde zender
150 kHz tot 80 MHz
80 MHz tot 800 MHz
800 MHz tot 2,5 GHz
d = 1.2 P
d = 1.2 P
d = 1.2 P
W
0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
Voor zenders met een ander nominaal maximumvermogen dan hierboven vermeld, kan de aanbevolen
scheidingsafstand (d) in meter (m) worden geschat met behulp van de vergelijking die van toepassing is op de frequentie
van de zender, waarbij P staat voor het nominale maximaal uitgangsvermogen van de zender in watt (W) volgens defabrikant van de zender.
OPMERKING 1: Bij 80 MHz en 800 MHz geldt de scheidingsafstand voor het hogere frequentiebereik.
OPMERKING 2: Het kan zijn dat deze richtlijnen niet op alle situaties van toepassing zijn. Elektromagnetische propagatie
wordt beïnvloed door absorptie en reflectie van gebouwen, voorwerpen en mensen.

ClassificatieType bescherming tegen elektrische schokken:
Klasse II	Bescherming tegen elektrische schokken is			bereikt door dubbele isolatie.
Mate van bescherming tegen elektrische schokken:
	Type BF Apparatuur die een bepaalde		mate van bescherming biedt tegen elektrische		schokken met betrekking tot		1) toegestane lekstroom;		 2) betrouwbaarheid van aardaansluiting(indien aanwezig).
		Niet bestemd voor rechtstreekse harttoepassingen.
Onafhankelijke testen voor de norm voor medischeelektrische apparatuur:	Conform bevonden aan• IEC 60601-1 uitgave 3.1: Medische elektrische
apparatuur - Deel 1: Algemene veiligheidseisen.
• CAN/CSA C22.2 nr. 60601-1-8 uitgave 2.1,
Medische elektrische toestellen - Deel 1:
Algemene veiligheidseisen.
Bescherming tegen potentiële elektromagnetische ofandere interferentie tussen de apparatuur en andere
toestellen.
• Getest en conform bevonden aan EN 60601-1-2/IEC 60601-1-2: 4e uitgave.

	• Conform bevonden aan RTCA/DO160 deel 21	categorie M.
• CISPR 11/EN 55011 Klasse B Groep 1,
	‘HF-apparatuur voor industriële, wetenschappelijke
en medische doeleinden (zgn. ISM-apparatuur)’.
• FCC deel 15, subdeel B - klasse B,	Onbedoelde radiatoren.
IP22 - Bescherming tegen het binnendringen vanvingers of soortgelijke voorwerpen met een
diameter van meer dan 12,5 mm. Verticaal druipend
water heeft geen schadelijke gevolgen wanneer debehuizing onder een hoek van 15° wordt gekanteld
ten opzichte van de normale positie ervan.
Toegestane methode voor reiniging en infectiebeheersing: raadpleeg het deel ‘Reiniging, zorg en
routineonderhoud’ van deze gebruikershandleiding
van de FreeStyle.
Mate van veiligheid van de toepassing in deaanwezigheid van brandbare anesthesiegassen:	De apparatuur is niet geschikt voor een dergelijke
toepassing.
Bedrijfswijze: continubedrijf.

PN MN232-C4 L | Gebruikershandleiding NL - 239

FreeStyle Comfort
Brugerkontrolelementer og indikatorer for systemstatus
ISO 7000
Brugervejledning; brugsanvisning.
Reg.nr. 1641
Land og fremstillingsdato. "CC"er den to-bogstavs landekode
for fremstillingslandet.
Fremstillingsdatoen er i formatetÅÅÅÅ-MM-DD. Reg.nr. 6049

ISO 7010: Grafiske symboler – sikkerhedsfarverog -skiltning – registreret sikkerhedsskiltning
Brugervejledningen skal læses.
Reg.nr. M002
Må ikke udsættes for åben ild, brandeller gnister. Åben antændelseskildeog rygning er ikke tilladt.
Reg.nr. P003
Der må ikke ryges i nærheden af
enheden eller under drift.
Reg.nr. P002

Må ikke udsættes for regn.
Opbevares tørt. Reg.nr. 0626
Stablegrænse efter antal, hvor "n"angiver det maksimalt tilladte antalenheder. Reg.nr. 2403
Katalognummer. Reg.nr. 2493

Advarsel! Reg.nr. W001
IEC 60417
Type BF-del (beskyttelsesgrad
mod elektrisk stød). Reg.nr. 5333

Serienummer. Reg.nr. 2498
Område for opbevarings- eller
driftstemperatur. Reg.nr. 0632
Område for opbevaringsfugtighed.
Reg.nr. 2620
Atmosfærisk trykbegrænsning.
Reg.nr. 2621
Producentens navn og adresse.
Reg.nr. 3082

Klasse II-udstyr, dobbeltisoleret.
Reg.nr. 5172
EU-forordning 2017/745
Autoriseret repræsentant i EU
Hvis mærkaten med produktets
unikke enhedsidentifikator (UDI) har
symbolet CE####, opfylder enheden
kravene vedrørende medicinsk udstyr.
Symbolet CE#### angiver nummerpå bemyndiget organ.
Yderligere symboler

Denne side op. Reg.nr. 0623

Må ikke anbringes i nærheden afbrændbare materialer, olie og fedt.

Skrøbelig – skal håndteres forsigtigt.
Reg.nr. 0621

Må ikke adskilles.

Indeholder farlige stoffer.
Reg.nr. 3723

Øg hastigheden.

Importør. Reg.nr. 3725

Reducer hastigheden.

O2-udtag

Knappen Menu.
Afbryder. Reg.nr. 5009
RTCA/DO160, afsnit
21, i overensstemmelse med
kategori M

240 - DK PN MN232-C4 L | Brugermanual

Producenten af denne POC harfastslået, at denne enhed er ioverensstemmelse med alle gældende
FAA-acceptkriterier og brug om bordpå fly.

FreeStyle Comfort
Producenten af denne POC har fastslået,at denne enhed er i overensstemmelse
med alle gældende FAA-acceptkriterierog brug om bord på fly.

EU Rådets direktiv 2012/19/EU
WEEE
Dette symbol har til formål at minde
ejere af udstyr om at returnere det
til et genvindingsanlæg, når det er
udtjent, i henhold til direktivet om
affald af elektrisk og elektronisk
udstyr (WEEE). Vores produkter
overholder begrænsningerne i
RoHS-direktivet. De indeholder
ikke mere end spor af bly eller
andre farlige stoffer.

Genbrugssymbol
Autoriseret repræsentant i Schweiz(ADD)
Hvis enheden bærer UKCA-mærket
som vist med UKCA####, derangiver det bemyndigede organs
nummer, overholder denne enhed
UKCA-reglerne.
21 CFR 801.15
Ifølge føderal amerikansk lovgivningmå dette apparat kun sælges af enlæge eller efter ordination af en læge.

IEC 60601-1-8
Teknisk alarm med lav prioritet
ISO 15223-1
Medicinsk udstyr. Reg.nr. 5.7.7

ETL-certificering
Certificeret til det amerikanskeog canadiske marked i henhold til
amerikanske og canadiske standarder.
Indeholder FCC ID: WAP2001
FCC-erklæring
Denne enhed kan indeholde CYBLE-022001-00,
herunder antennen 2450AT18B100 fra Johnson
Technology – overholder del 15 af FCCreglerne. Denne enhed opfylder kravene formodulsender godkendt i henhold til FCC's
offentlige erklæring DA00-1407. Betjening af
senderen er underlagt følgende to betingelser: (1)Enheden må ikke forårsage skadelig interferens,og (2) denne enhed skal acceptere al interferens,
den måtte modtage, herunder interferens der
kan påvirke betjeningen negativt. Produktet er
muligvis omfattet af ét eller flere amerikanske
eller internationale patenter. Gå til vores websted
nedenfor for at se en liste over gældende patenter.
Pat.: www.caireinc.com/corporate/patents

Unikt enheds-id-nr. 5.7.10
A S/NZS 4417.1, AS/NZS 4417.2
ACMA Regulatory ComplianceMark – angiver overholdelseaf ACMA-regulativer (ADD)Trådløs teknologi
Tilslutningsmuligheder

SSP, iAP2, GATT

Frekvens

2402 til 2480 MHz

Maks. RF-strømudgang+4 dBmDriftsområde

10 m (klasse 2)

Det anbefales, at enheden befinder sig mindst
0,43 tommer (1,1 cm) fra kroppen under drift.
Dette gælder ikke masker, luftslanger eller tilbehør.

Grafiske symboler til brug på IEC 60417-udstyrJævnstrøm. Reg.nr. 5031
Vekselstrøm. Reg.nr. 5032
IEC 60601-1
Dryptæt

PN MN232-C4 L | Brugermanual DK - 241

FreeStyle Comfort
Udpakning af din FreeStyle Comfort
Kontrollér, at alle dele,der er anført, er i pakken.
Kontakt omgående
iltudstyrsleverandøren,hvis noget mangler.

Standardtilbehør• Bærbar iltkoncentrator• Bæretaske• Skulderrem• 8-cellet batteripakke• Vekselstrømsforsyning• Vekselstrømskabel• Jævnstrømforsyning• Brugermanual

Valgfrit ekstraudstyr• Ekstra 8-cellet batteripakke• 16-cellet batteripakke• Ekstern oplader• TilbehørstaskeLær din FreeStyle Comfort at kendeSæt dig ind i nøglefunktionerne i FreeStyle Comfort og dens brugergrænseflade.

Set forfra
Remfastgørelse

Brugergrænseflade

IltudtagLuftudtag

Luftindtag/filtre
Luftindtag/filtre

Tap på
strømstik
Batteripakke
Udløserknap til
batteripakke
242 - DK PN MN232-C4 L | BrugermanualFreeStyle Comfort
Set ovenfra
DisplayØg hastighedenTænd/sluk-knap

Knappen Menu

Reducer hastigheden

Display på batteristrøm
Flowindstilling

Batterivisning

Indikatoren autoSAT
Alarmdisplay

Resterendebatteriprocent

Indikator forlevering af bolus

Display på vekselstrøm/jævnstrøm

Flowindstilling
BatterivisningOmråde til
alarmvisning

Opladningsindikator

PN MN232-C4 L | Brugermanual DK - 243

FreeStyle Comfort
Hvad er en iltkoncentrator?
Den luft, vi indånder, indeholder ca. 21 % ilt,
78 % kvælstof og 1 % andre luftarter. I FreeStyle®
Comfort® suges almindelig luft ind i maskinen
igennem luftindtagene. Derefter strømmer den
igennem et adsorberende materiale, der kaldes
en molekylesi. Dette materiale adskiller ilten frakvælstoffet og lader kun ilten trænge igennem.
Resultatet er en strøm af meget ren ilt, der leveres
til brugeren.
FreeStyle Comfort kombinerer avanceret
iltkoncentratorteknologi med iltbevarende teknologi
i en bærbar enhed, der sikrer den bedst mulige
mobilitet og ambulante brug. Den avanceredepulserende flowlevering registrerer hurtigt, nårbrugeren ånder ind, og leverer en iltpuls i starten af
hver indånding. Pulserende flowlevering er ideelt til
aktiviteter og brug uden for hjemmet. I modsætning
til et uafbrudt flow bevarer den ilt og forlænger
batteriets levetid. FreeStyle Comfort er let nok
til at kunne bæres med en skulderrem. Produktet
kan køre på vekselstrøm (stikkontakt), jævnstrøm(i et køretøj) eller på batteri.

Tilsigtet anvendelse
CAIRE FreeStyle Comfort-iltkoncentrator er
beregnet til administration af supplerende ilt.
Enheden er ikke beregnet som respirator, og den
har ingen yderligere patientovervågningsfunktioner.

Grunden til, at din læge harordineret ilt
Mange mennesker lider af en række forskellige
hjerte-, lunge- og andre luftvejssygdomme.
Et væsentligt antal af disse personer kan havefordel af supplerende iltbehandling i hjemmet, pårejser eller ved deltagelse i daglige aktiviteter uden
for hjemmet.
Ilt er en luftart, der udgør 21 % af den luft, viindånder, og kroppen er afhængig af en konstantiltforsyning for at kunne fungere ordentligt.
Din læge har ordineret en iltstrøm eller en
indstilling, der passer til netop din tilstand.

244 - DK PN MN232-C4 L | Brugermanual

Selvom ilt ikke er et vanedannende lægemiddel,
kan uautoriseret iltbehandling være farlig. Du skalrådføre dig med en læge, inden du anvender denne
iltkoncentrator. Leverandøren, der leverer dit
iltudstyr, vil vise dig, hvordan du skal indstille den
ordinerede flowhastighed.

Brugerprofil
Koncentratorer er beregnet til at levere supplerende
ilt til brugere, der oplever ubehag som følge
af lidelser, der nedsætter lungernes evne til attransportere ilt fra luften til blodet. Bærbare
iltkoncentratorer (POC'er) hverken indeholder
eller lagrer ilt. De skal ikke genopfyldes, og de
kan genoplades, hvor som helst du har adgang tiljævnstrøm eller vekselstrøm (undtagen ombord
på fly). Det gør brugeren relativt selvforsynende
med hensyn til hjemmebrug, ambulant brug(både i og uden for hjemmet), mobilitet og generel
livsstil. Brug af iltkoncentratoren kræver ordination
fra lægen.
Den typiske bruger er ældre end 65 år og lider af enrække forskellige luftvejssygdomme, bl.a. kronisk
obstruktiv lungesygdom (KOL). Brugerne hartypisk gode kognitive evner og skal være i standtil at kommunikere ubehag. Hvis brugeren ikke
er i stand til at kommunikere ubehag eller læseog forstå mærkaterne på iltkoncentratoren og
brugervejledningen, anbefales brug kun underovervågning af en person, som kan dette. Hvis der
opleves ubehag under brug af koncentratoren,
anbefales det, at brugerne kontakter hjemmeplejen.
Det anbefales også, at brugerne har ekstra ilttilgængelig (dvs. iltflaske) i tilfælde af strømsvigt
eller fejl i koncentratoren. Der kræves ingensærlige færdigheder eller brugerevner for at bruge
koncentratoren.

FreeStyle Comfort
Vigtigt!
Sikkerhedsinstruktionerne er defineret som følger:
ADVARSEL: VIGTIGE SIKKERHEDSOPLYSNINGER I FORBINDELSE MED RISIKO FOR ALVORLIG PERSONSKADE.
FORSIGTIG: Vigtige oplysninger
om forebyggelse af skader
på FreeStyle Comfort.
Bemærk: Oplysninger, der kræver særlig
opmærksomhed.

Indikationer for anvendelse
ADVARSEL: DET ER MEGET VIGTIGTAT VÆLGE DET ORDINEREDE ILTNIVEAU. UNDGÅ AT ÆNDRE FLOWINDSTILLINGEN, MEDMINDRE EN AUTORISERET KLINISK ARBEJDENDE LÆGEHAR BEDT DIG OM AT GØRE DET.
ADVARSEL: DENNE ENHED ER IKKE BLEVETTESTET ELLER GODKENDT TIL BRUG MED
PÆDIATRISKE PATIENTER.
ADVARSEL: IFØLGE NATIONAL LOVGIVNINGI USA MÅ DENNE ENHED KUN SÆLGESELLER UDLEJES EFTER HENVISNING FRAEN LÆGE ELLER ANDET AUTORISERETSUNDHEDSPERSONALE.
ADVARSEL: DENNE ENHED MÅ IKKE ANVENDESSOM RESPIRATOR. YDERLIGERE OVERVÅGNING
KAN VÆRE PÅKRÆVET FOR BRUGERE, DER
IKKE ER I STAND TIL AT KOMMUNIKERE UBEHAG
UNDER BRUG AF DENNE ENHED. BRUGERE MED
HØRE- OG/ELLER SYNSNEDSÆTTELSE SKAL HAVEASSISTANCE MED OVERVÅGNING AF ALARMER.
ADVARSEL: INDSTILLINGER FOR
PULSDOSISTILSTAND (PULSE DOSE MODE)
SKAL FASTSÆTTES FOR HVER ENKELT BRUGERI HENHOLD TIL VEDKOMMENDES BEHOV UNDERHVILE, TRÆNING OG PÅ REJSER.

ADVARSEL: NOGLE AF PATIENTENSVEJRTRÆKNINGER UDLØSER MULIGVISIKKE DEN ILTBESPARENDE FUNKTION.
ADVARSEL: FREESTYLE COMFORT ER
IKKE BEREGNET TIL BRUG SAMMEN MED
TRAKEOTOMEREDE PATIENTER.
ADVARSEL: GRAVIDE OG AMMENDE KVINDERBØR IKKE BRUGE TILBEHØR, DER ANBEFALESI DENNE VEJLEDNING, DA DISSE KAN
INDEHOLDE FTALATER.

Kontraindikationer for anvendelse
ADVARSEL: UNDER VISSE FORHOLDKAN DET VÆRE FARLIGT AT ANVENDE
IKKE-ORDINERET ILT. ENHEDEN BØR
KUN ANVENDES EFTER ORDINATIONFRA EN LÆGE.
ADVARSEL: DENNE ENHED ER IKKE BEREGNETTIL BRUG I ELLER I NÆRHEDEN AF MAGNETISKRESONANSMILJØER, SÅSOM MR-MASKINER.
ADVARSEL: MÅ IKKE ANVENDES I NÆRHEDEN
AF BRÆNDBARE ANÆSTESIMIDLER.
ADVARSEL: SOM MED ALLE ELEKTRISKEENHEDER KAN BRUGEREN OPLEVEPERIODER UDEN DRIFT SOM FØLGE AFSTRØMAFBRYDELSER ELLER BEHOV FOR,AT ILTKONCENTRATOREN SERVICERES AF ENKVALIFICERET TEKNIKER. ILTKONCENTRATORENER IKKE VELEGNET TIL BRUGERE, DER VILLE
FÅ ALVORLIGE SUNDHEDSMÆSSIGE FØLGER
AF EN SÅDAN MIDLERTIDIG STRØMAFBRYDELSE.
ADVARSEL: DENNE ENHED ER IKKE GODKENDTTIL BRUG I ALLE MILJØER. EKSPONERING FOR
VISSE ELEKTROMAGNETISKE KILDER KAN
RESULTERE I FORSTYRRELSER ELLER UTILSIGTETENHEDSYDELSE. I TILFÆLDE AF FORSTYRRELSERSKAL ENHEDEN SLUKKES STRAKS.

ADVARSEL: DENNE ENHED ER BEREGNET TIL
BRUG I HJEMMESUNDHEDSPLEJE OG LIGNENDEAMBULANTE MILJØER.
ADVARSEL: DENNE ENHED ER BEREGNET TIL AT
BLIVE BRUGT AF DEN PATIENT, DER MODTAGEROXYGENTERAPI.

PN MN232-C4 L | Brugermanual DK - 245

FreeStyle Comfort
Sikkerhedsretningslinjer
ADVARSEL: DET ER IKKE TILLADTAT MODIFICERE UDSTYRET.
ADVARSEL: DENNE ENHED LEVERER EN HØJ
ILTKONCENTRATION, DER FREMMER HURTIGFORBRÆNDING. RYGNING ELLER BRUG AF ÅBENILD ER IKKE TILLADT I SAMME RUM SOM (1)DENNE ENHED ELLER (2) ILTBÆRENDE UDSTYR.
MANGLENDE OVERHOLDELSE AF DENNE
ADVARSEL KAN MEDFØRE BRAND, MATERIELSKADE OG/ELLER FYSISKE SKADER ELLERDØDSFALD.
ADVARSEL: ANVEND IKKE APPARATET I ET
AFLUKKET ELLER BEGRÆNSET RUM (DVS.
ET LILLE ETUI ELLER EN LILLE TASKE),
HVOR VENTILATIONEN KAN VÆRE DÅRLIG.
DET KAN MEDFØRE OVEROPHEDNING AF
ILTKONCENTRATOREN, HVILKET FORRINGERYDEEVNEN.
ADVARSEL: KONCENTRATOREN SKAL
PLACERES PÅ GOD AFSTAND AF RØG,FORURENENDE STOFFER ELLER GASSER.
ADVARSEL: I TILFÆLDE AF EN ALVORLIGHÆNDELSE MED DENNE ENHED SKAL
BRUGEREN STRAKS INDBERETTE HÆNDELSEN
TIL LEVERANDØREN OG/ELLER PRODUCENTEN.
EN ALVORLIG HÆNDELSE DEFINERES SOM ENSKADE, DØDSFALD ELLER POTENTIALE TIL
AT FORÅRSAGE PERSONSKADE/DØDSFALD,HVIS HÆNDELSEN GENTAGER SIG. BRUGERENKAN OGSÅ INDBERETTE HÆNDELSEN TIL DEN
KOMPETENTE MYNDIGHED I DET LAND, HVOR
HÆNDELSEN OPSTOD.
ADVARSEL: BRUGEN AF VISSE TYPERILTADMINISTRATIONSUDSTYR, DER IKKEER SPECIFICERET TIL BRUG MED DENNE
ILTKONCENTRATOR, KAN PÅVIRKEYDEEVNEN. ANBEFALET TILBEHØR ER
ANGIVET I NÆRVÆRENDE VEJLEDNING.
ADVARSEL: HVIS ILTKONCENTRATOREN
BLIVER TABT, BESKADIGET ELLERUDSÆTTES FOR VAND, SKAL DU KONTAKTE
LEVERANDØREN FOR AT FÅ DEN UNDERSØGT
OG EVENTUELT REPARERET. UNDGÅ AT BRUGE
ILTKONCENTRATOREN, HVIS LEDNINGEN ELLERSTIKKET ER BESKADIGET.
ADVARSEL: SØRG FOR AT SIKRE,FASTSPÆNDE OG PÅ ANDEN MÅDE FASTGØRE
ILTKONCENTRATOREN UNDER KØRSEL FOR AT
FORHINDRE BESKADIGELSE ELLER ULYKKER.
246 - DK PN MN232-C4 L | Brugermanual

ADVARSEL: BESTEMMELSER FRA
DET AMERIKANSKE DEPARTMENT OF
TRANSPORTATION (DOT) OG DE FORENEDE
NATIONER (FN) KRÆVER, AT BATTERIETFJERNES FRA ENHEDEN I FORBINDELSE
MED INTERNATIONALE FLYVNINGER, HVOR
ILTKONCENTRATOREN ER CHECKET IND
SOM BAGGAGE. VED TRANSPORT AF
ILTKONCENTRATOREN SKAL BATTERIET OGSÅFJERNES FRA ENHEDEN OG EMBALLERESSEPARAT.
ADVARSEL: UNDGÅ AT BRUGE UDSTYRET
UDEN FØRST AT HAVE LÆST OG FORSTÅET
DENNE VEJLEDNING. HVIS DU IKKE FORSTÅRADVARSLERNE OG INSTRUKTIONERNE, SKAL
DU KONTAKTE LEVERANDØREN AF UDSTYRET,FØR DU FORSØGER AT BRUGE DET. I MODSATFALD KAN DET MEDFØRE PERSON- ELLER
TINGSSKADE.
ADVARSEL: HVIS DU FØLER DIG UTILPAS ELLER
BEFINDER DIG I EN MEDICINSK NØDSITUATION,SKAL DU STRAKS SØGE LÆGE.
ADVARSEL: HVIS ILTKONCENTRATOREN
BENYTTES UDEN FOR DE ANGIVNE
DRIFTSTEMPERATURER, KAN DET BEGRÆNSE
KONCENTRATORENS EVNE TIL AT OPFYLDE
SPECIFIKATIONERNE FOR ILTKONCENTRATION.
DU KAN SE TEMPERATURGRÆNSERNEI AFSNITTET OM SPECIFIKATIONER I DENNE
VEJLEDNING.
ADVARSEL: UNDGÅ AT BENYTTE OLIE,
FEDTSTOF ELLER PETROLEUMSBASEREDE
ELLER ANDRE BRÆNDBARE PRODUKTERSAMMEN MED ILTBÆRENDE TILBEHØR ELLER
ILTKONCENTRATOREN. KUN VANDBASEREDEPRODUKTER, ILTKOMPATIBLE LOTIONER ELLERSALVER BØR ANVENDES. ILT FREMSKYNDERFORBRÆNDING AF BRÆNDBARE STOFFER.
ADVARSEL: FORKERT BRUG AF BATTERIETKAN FÅ BATTERIET TIL AT BLIVE VARMT ELLERANTÆNDE OG KAN FORÅRSAGE ALVORLIGPERSONSKADE. UNDGÅ AT PRIKKE HUL I, SLÅ PÅ,TRÆDE PÅ ELLER TABE BATTERIET ELLER PÅ
ANDEN MÅDE UDSÆTTE BATTERIET FOR STØDELLER SLAG. BRUG AF ET BESKADIGET BATTERIKAN FORÅRSAGE ALVORLIG PERSONSKADE.
ADVARSEL: BATTERIETS METALKONTAKTER MÅ
IKKE KORTSLUTTES MED METALGENSTANDE,SOM F.EKS. NØGLER ELLER MØNTER. DETTE
KAN MEDFØRE, AT DER GENERERES GNISTER
ELLER OVERDREVEN VARME.

FreeStyle Comfort
ADVARSEL: HVIS DEN BÆRBARE
ILTKONCENTRATOR BENYTTES UDENDØRS
SAMMEN MED JÆVNSTRØMFORSYNINGEN,MÅ STRØMFORSYNINGEN KUN TILSLUTTESET GFI-UDTAG (GROUND FAULT INTERRUPTED).
ADVARSEL: ILTKONCENTRATOREN BØR IKKEBRUGES VED SIDEN AF ELLER STABLET MED
ANDET UDSTYR. HVIS DET ER NØDVENDIGTAT ANVENDE DEN PLACERET VED SIDEN
AF ELLER STABLET, SKAL ENHEDENOVERVÅGES FOR AT SIKRE KORREKT DRIFT.
ADVARSEL: BRUG AF ANDRE KABLER OG
ADAPTERE END DEM, DER ER ANGIVET, MED
UNDTAGELSE AF KABLER OG ADAPTERE, SOM
PRODUCENTEN AF DEN ELEKTRISKE MEDICINSKEANORDNING SÆLGER SOM RESERVEDELETIL INTERNE KOMPONENTER, KAN MEDFØRE
FORØGET EMISSION AF REDUCERET IMMUNITETFRA ILTKONCENTRATOREN.
ADVARSEL: BRUG KUN ELEKTRISK SPÆNDING
SOM ANGIVET PÅ ENHEDENS MÆRKAT.
ADVARSEL: BRUG IKKE FORLÆNGERLEDNINGER
TIL DENNE ENHED, OG TILSLUT IKKE FOR MANGESTIK I DEN SAMME STIKKONTAKT. BRUGEN
AF F​ ORLÆNGERLEDNINGER KAN PÅVIRKEENHEDENS YDEEVNE. ALT FOR MANGE STIKI DEN SAMME STIKKONTAKT KAN RESULTERE
I EN OVERBELASTNING PÅ EL-PANELET,
SÅ AFBRYDEREN/SIKRINGEN AKTIVERESELLER GÅR I BRAND, HVIS AFBRYDEREN ELLERSIKRINGEN IKKE FUNGERER.
FORSIGTIG: I tilfælde af en alarm, ellerhvis det konstateres, at iltkoncentratoren
ikke fungerer korrekt, henvises der til
afsnittet Fejlfinding i denne vejledning.
Kontakt udstyrsleverandøren, hvis
problemet ikke kan løses.

ADVARSEL: UNDGÅ AT UDFØRE
ANDEN VEDLIGEHOLDELSE END
ANGIVET I DENNE VEJLEDNING.
ADVARSEL: KUN TILBEHØR, DER ER ANBEFALET
AF PRODUCENTEN. ENHVER ANDEN BRUG
KAN VÆRE FARLIG, MEDFØRE ALVORLIGE
SKADER PÅ ILTKONCENTRATOREN OG OPHÆVEGARANTIENS DÆKNING.
ADVARSEL: PRODUCENTEN ANBEFALER, AT DU
HAR EN ALTERNATIV KILDE TIL SUPPLERENDE ILTI TILFÆLDE AF STRØMUDFALD, ALARMTILSTAND
ELLER MEKANISK FEJL. SPØRG DIN LÆGEELLER UDSTYRSLEVERANDØR, HVILKEN TYPERESERVESYSTEM DER ER PÅKRÆVET.
ADVARSEL: SØRG FOR, AT LUFTINDTAGETOG LUFTDYSERNE IKKE BLIVER BLOKERET.
UNDGÅ AT TABE ELLER INDSÆTTE GENSTANDEI ENHEDENS ÅBNINGER. DET KAN MEDFØREOVEROPHEDNING AF ILTKONCENTRATOREN,HVILKET FORRINGER YDEEVNEN.
ADVARSEL: RYGNING UNDER BRUG AF ILT ER
DEN STØRSTE ÅRSAG TIL BRANDULYKKER OG
RELATEREDE DØDSFALD. DU SKAL OVERHOLDEFØLGENDE SIKKERHEDSADVARSLER:ADVARSEL: UNDGÅ RYGNING, LEVENDE LYS
ELLER ÅBEN ILD I SAMME RUM SOM ENHEDENELLER ILTBÆRENDE TILBEHØR.
ADVARSEL: RYGNING UNDER BRUG AFNÆSEKANYLE KAN UDLØSE FORBRÆNDINGI ANSIGTET OG RISIKO FOR DØDSFALD.
ADVARSEL: HVIS DU RYGER, KAN FØLGENDE
3 TRIN MULIGVIS REDDE DIT LIV: SLUKILTKONCENTRATOREN, TAG KANYLEN AF,
OG FORLAD DET LOKALE, HVOR ENHEDENER PLACERET.

FORSIGTIG: Følg alle producentens anvisninger
for ikke at ophæve garantien.

PN MN232-C4 L | Brugermanual DK - 247

FreeStyle Comfort
ADVARSEL: DER SKAL VÆRE TYDELIG SKILTNING
MED "RYGNING FORBUDT – ILTAPPARAT I BRUG"I HJEM, HVOR DER ANVENDES ILTAPPARATER.
BRUGERE OG PLEJERE SKAL INFORMERES OM
FARERNE VED AT RYGE I NÆRHEDEN AF, ELLER
UNDER BRUG AF, MEDICINSK ORDINERET ILT.
ADVARSEL: UNDGÅ AT BENYTTE
ILTKONCENTRATOREN I NÆRHEDEN AF
BRÆNDBARE GASSER. DET KAN MEDFØRE
BRAND OG MEDFØLGENDE RISIKO FORTINGSSKADE, PERSONSKADE OG DØDSFALD.
ADVARSEL: HVIS DU TAGER NÆSEKANYLEN
AF OG ANBRINGER DEN PÅ TØJ, SENGETØJ,MØBLER ELLER ANDET POLSTRET MATERIALE,UDLØSES EN EKSPLOSIONSAGTIG BRAND, HVISDEN KOMMER I FORBINDELSE MED CIGARETTER,VARMEKILDER ELLER FLAMMER.
ADVARSEL: UNDGÅ AT EFTERLADE
NÆSEKANYLEN PÅ ELLER UNDER TØJ,SENGETØJ ELLER PUDER. HVIS ENHEDENTÆNDES UDEN AT VÆRE I BRUG, KAN ILTENGØRE MATERIALET BRÆNDBART. SØRG FOR,AT DER ER SLUKKET FOR STRØMMEN, NÅR
ILTKONCENTRATOREN IKKE ER I BRUG.
ADVARSEL: GODKENDT TILBEHØR TIL ILTTILFØRSEL ER DESIGNET TIL AT REDUCERE UDBREDELSEN AF BRAND AF HENSYN TIL PATIENTENSSIKKERHED. BRUG AF IKKE-GODKENDT TILBEHØR KAN ØGE RISIKOEN FOR PERSONSKADE
ELLER SKADE I FORBINDELSE MED BRAND.
ADVARSEL: ILTKONCENTRATOREN SKAL VÆRESLUKKET OG KOBLET FRA DEN EKSTERNE
STRØMFORSYNING, INDEN BATTERIPAKKENTAGES UD.
ADVARSEL: SMÅ DELE KAN UDGØRE
KVÆLNINGSFARE.
ADVARSEL: UNDGÅ KONTAKT MED UDLUFTNINGSHULLERNE I MERE END 10 SEKUNDER,DA DE KAN VÆRE VARME.
ADVARSEL: VIND OG KRAFTIG TRÆK KAN
HAVE NEGATIV INDFLYDELSE PÅ NØJAGTIGILTBEHANDLING.
ADVARSEL: DETTE PRODUKT KAN UDSÆTTE
DIG FOR KEMIKALIER, HERUNDER NIKKEL,
SOM I STATEN CALIFORNIEN ER KENDT FOR AT
FORÅRSAGE KRÆFT. FÅ FLERE OPLYSNINGER
PÅ WWW.P65WARNINGS.CA.GOV.
FORSIGTIG: Tag altid vekselstrømsforsyningen ud af stikkontakten,
før du kobler vekselstrømsforsyningen
fra iltkoncentratoren.
248 - DK PN MN232-C4 L | Brugermanual

FORSIGTIG: Iltkoncentratoren må ikke placeres,så det er vanskeligt at komme til strømkablet.
FORSIGTIG: Anbring altid iltslanger og ledninger
på en sådan måde, at man ikke kan falde i dem eller
blive kvalt af dem.
FORSIGTIG: Hvis iltkoncentratoren benyttes
i en bil, båd eller på andre jævnstrømkilder medjævnstrømforsyningen, skal du sørge for, at
kilden kører, før du tilslutter iltkoncentratoren.
Hvis disse instruktioner ikke overholdes, kan
det medføre, at strømforsyningen ikke kan tilføre
iltkoncentratoren strøm.
FORSIGTIG: Hvis den bil, hvori du bruger
iltkoncentratoren, slukkes, skal du frakobleog fjerne enheden fra bilen. Undgå at opbevare
iltkoncentratoren i en meget varm eller kold bileller i andre lignende miljøer med høje eller lavetemperaturer.
Bemærk: Bærbart og mobilt RF-kommunikationsudstyr
kan påvirke elektrisk medicinsk udstyr.
FORSIGTIG: UNDGÅ at ladeiltkoncentratoren eller strømforsyningenvære tilsluttet i bilen, hvis tændingen er
slukket. Det kan opbruge bilens batteri.
FORSIGTIG: Hvis iltkoncentratoren anvendes
eller opbevares uden for dens normale
driftstemperaturområde og omgivelsestrykområde
(højde), kan det påvirke ydeevnen og reducere
batteriets driftstid og/eller øge batteriets
opladningstid. Se afsnittet Specifikationer
i denne manual for at få oplysninger om
temperaturgrænser for opbevaring og drift.
FORSIGTIG: Iltkoncentratorer med batterier:Opbevares et køligt og tørt sted for at sikre langbatterilevetid. Hvis du opbevarer iltkoncentratoren
over længere tid ved høje temperaturer eller med
et fuldt opladet eller et helt afladet batteri, kan det
forringe batteriets generelle holdbarhed. Forsøg
ikke at åbne batteriet. Der er ingen dele inde ibatteriet, som kan serviceres. Opbevar batterieruden for børns rækkevidde.
FORSIGTIG: BRUG KUN batterier fra producenten.
Kontakt udstyrsleverandøren eller kommunen med
henblik på korrekt bortskaffelse af batterier.

FreeStyle ComfortBemærk: For at afbøde skader forårsaget af svigt af
koncentrator eller strømsvigt bør udstyrsleverandøren
vurdere behovet for backup-ilt under installationen
og med jævne mellemrum, når følgende egenskaberændres:Bemærk: Hvis iltkoncentratoren har været
opbevaret i længere tid uden for dens normale
driftstemperaturområde, som er anført i
specifikationstabellen ovenfor, skal enheden førstopnå normal driftstemperatur, før den tændes.

- Patientens tilstandBemærk: Udstyrsleverandøren bør vurdere behovet
for backup-ilt i tilfælde af svigt af koncentratoren
eller strømsvigt under installationen og med jævne
mellemrum, når følgende egenskaber ændres:- Det miljø, som patienten lever i- Evnen til at forsyne patienten med
reserveforsyninger af supplerende ilt.
Bemærk: Sørg for, at kanylen er helt indsat og sidder
fast. Det sikrer, at iltkoncentratoren kan detektere
inspiration korrekt med henblik på iltlevering.
Under inhalering skal du kunne høre og føle iltflow
på nasalkanylen.

- Patientens tilstand- Det miljø, som patienten lever i- Evnen til at forsyne patienten med
reserveforsyninger af supplerende ilt.

Bemærk: Lithiumbatterier kan permanent miste
kapacitet, hvis de udsættes for kraftig varme, når deer enten helt opladet eller helt afladet. Ved længere
opbevaring anbefales det at oplade batterierne
25-50 % og opbevare dem i et temperaturområde
på -20 °C - 20 °C (-4 °F - 68 °F).
Bemærk: Undgå at bruge iltkoncentratoren, uden
at luftindtagsfilteret er isat. Hvis der er medleveret
et ekstra filter, skal reservefilteret isættes, før det
snavsede filter rengøres. Rengør det snavsede filter
med varmt vand og sæbe, og tør det grundigt før brug.
Bemærk: Producenten anbefaler ikke sterilisering
af udstyret.
Bemærk: Der er aldrig fare for at løbe tør for ilt
i et lokale, når du bruger iltkoncentratoren.
ADVARSEL: ILTKONCENTRATOREN MÅ
IKKE ANVENDES ELLER TILSLUTTESEKSTERNE STRØMFORSYNINGER(VEKSELSTRØM/JÆVNSTRØM),HVIS DER IKKE ER ISAT ET BATTERI.
GØR IKKE FORSØG PÅ AT TÆNDE
ENHEDEN ELLER SLUTTE DEN TIL
EN EKSTERN STRØMFORSYNING,
HVIS BATTERIPAKKEN ER TAGET UD.
BATTERIPAKKEN MÅ KUN TAGES UD, NÅR
ENHEDEN ER SLUKKET OG KOBLET FRA DENEKSTERNE STRØMFORSYNING. GØR IKKEFORSØG PÅ AT TAGE BATTERIPAKKEN UD, MENSENHEDEN ER TÆNDT ELLER TILSLUTTET ENEKSTERN STRØMFORSYNING.

PN MN232-C4 L | Brugermanual DK - 249

FreeStyle Comfort
Specifikationer
FreeStyle Comfort
Mål (med enkelt batteri)

10,0 x 7,3 x 3,1 tomme (25,4 x 18,5 x 7,9 cm)

Mål (med dobbelt batteri)

11,0 x 7,3 x 3,1 tomme (27,9 x 18,5 x 7,9 cm)

Vægt (med enkelt batteripakke)

2,3 kg (5 pound)Vægt (med dobbelt batteripakke)

2,7 kg (6 pound)

Nominelt støjniveau***

39,93 dB(A) ± 0,42 dB(A) @ 2

Flowindstillinger

Pulseringsindstilling 1, 2, 3, 4, 5

O2-udgangsindstilling([@20bpm STPD] Alle lydstyrker +/- 15 %)Iltkoncentration*
Iltmonitor

Indstilling 1 – 210 ml/min
Indstilling 2 – 420 ml/min
Indstilling 3 – 630 ml/min
Indstilling 4 – 840 ml/min
Indstilling 5 – 1050 ml/min
90 % (+5,5 % / -3 %)JaMaks. udgangstryk< 30 PSI

Doseringssensitivitet> -0,1 cm H2O (justerbar)

Vekselstrøm

100-240 V vekselstrøm, 50-60 Hz, 120 W maks.

Jævnstrøm

11-18 VDC (maks. 10 ampere)

Batteritype Lithium-ion
Batterikapacitet

Enkeltcelle: 6700 mAh, dobbeltcelle: 13400 mAh

Batterigenopladningstid

Enkelt batteri: 3,5 timer
Dobbelt batteri: 6,0 timer

Batteripakkens driftstid

Varighed af enkelt batteri: Indstilling 2 – op til 4 timer
Varighed af dobbelt batteri: Indstilling 2 – op til 8 timer

Driftstemperatur**

5 °C til 40 °C (41 °F til 104 °F)

Transport- og opbevaringstemp. (Enhed)-25 °C til +70 °C (-13 °F til +158 °F) op til 3048 m (10000 ft)

Transport- og opbevaringstemp. ri-25 °C til +70 °C (-13 °F til +158 °F) op til 3048 m (10000 ft)

Driftshøjde**
Driftsluftfugtighed
Fugtighed ved opbevaringBegrænset garanti-381 til 3048 m (-1250 til 10000 ft) (testet til 700-1060 hPa)
15-95 % relativ fugtighed (ikke-kondenserende)0-90 % ikke-kondenserende
Se garantierklæring

* Baseret på et atmosfærisk tryk på 14,7 psi (101 kPa) ved 21 °C (70 °F)** Enheden vil fungere i alle miljøer, så længe omgivelserne er inden for de specificerede temperatur- og
fugtighedsområder. Brug uden for disse driftsspecifikationer kan begrænse koncentratorens evne til at opfylde
specifikationerne for iltkoncentration ved højere flowhastigheder. Opvarmningsskærmbilledet vises muligvis, indtil
koncentratoren har nået målkoncentrationen eller i op til 15 minutter. Bliv ved med at bruge enheden i denne periode.
*** Lydniveau målt pr. testmetode nr. 14-1 10/2018 MDS-Hi.
Den forventede levetid er minimum fem år.

250 - DK PN MN232-C4 L | BrugermanualFreeStyle Comfort
Brugsanvisning
1. Anbring og opstil FreeStyle Comfort et sted,
der er egnet til anvendelsen.
a. Koncentratoren skal være godt ventileret,således at luftindtagsfiltrene og luftudtaget
ikke blokeres.
b. Koncentratoren skal opstilles, så det er
nemt at høre og se alle hørbare og synlige
indikatorer og alarmer.
c. Sørg for, at luftindtagsfiltrene er monteret,inden enheden tages i brug. Hvis der
mangler et filter i enheden, skal du kontakte
leverandøren af udstyret.
2. Sæt en batteripakke i koncentratoren.
a. Ret batteriet ind efter bunden af huset
fra midten af FreeStyle Comfort.

3. Sæt veksel- eller jævnstrømforsyningen istrømstikket nederst til højre foran på enheden.

4. Slut en næsekanyle til iltudtagsstudsen.
Sørg for, at den er tilsluttet korrekt inden brug.
BEMÆRK: Brug af andre kanyler end den anbefaledekan påvirke ydeevnen.

Bemærk: Til udstyrsleverandøren: Følgende
iltadministrationstilbehør anbefales til brug sammen
med FreeStyle Comfort:b. Før batteriet indad, indtil du hører et klik,
som bekræfter, at det er låst fast.

• Næsekanyle med 2,1 m (7 fod) slange (maks. l/min.:
6 l/min.): CAIRE varenummer 5408-SEQ• Yderligere information om anbefalet tilbehør
er tilgængeligt online på www.caireinc.com.
5. Tænd for FreeStyle Comfort ved at trykke påtænd/sluk-knappen, og vælg den foreskrevne
flowindstilling ved hjælp af knappen til øgning(+) eller reduktion (-) af flow.

PN MN232-C4 L | Brugermanual DK - 251

FreeStyle Comfort
Respirationsfrekvens
15
20
25
30
35
40

Der udsendes et kortvarigt lydsignal,
som angiver, at enheden er tændt.
Bemærk: Når FreeStyle Comfort tændes første gang,er standardflowindstillingen 2. Hvis indstillingen er blevetændret, tændes enheden ved den forrige flowindstilling.
6. Sæt næsekanylen på ansigtet i henhold til
producentens instruktioner, som følger med
kanylen, og træk vejret helt normalt. Du bør
kunne mærke, at et pust af ilt leveres igennemnæsekanylen, hver gang du ånder ind.
7. Hvis du vil ændre flowindstillingen, skal du
trykke på knappen til øgning (+) eller reduktion(-) for at indstille den ønskede flowhastighed.
Det er normalt at høre forskellige lyde, nårflowindstillingerne ændres.

Respirationsfrekvens
15
20
25
30
35
40

Indstilling for autoSAT-pulsvolumen (ml)*
1
2
3
4
5
10,5
10,5
10,5
10,5
10,5
10,5

21,0
21,0
21,0
21,0
21,0
21,0

31,5
31,5
31,5
31,5
30,0
26,3

*Tolerancen er +/- 15 %252 - DK PN MN232-C4 L | Brugermanual

42,0
42,0
42,0
35,0
30,0
26,3

52,5
52,5
42,0
35,0
30,0
26,3

1
14,0
10,5
8,4
7,0
6,0
5,3

Indstilling for
ikke-autoSAT-pulsvolumen (ml)*
2
3
4
5
28,0
21,0
16,8
14,0
12,0
10,5

42,0
31,5
25,2
21,0
18,0
15,8

56,0
42,0
33,6
28,0
24,0
21,0

70,0
52,5
42,0
35,0
30,0
26,3

8. Sluk for FreeStyle Comfort ved at trykke påtænd/sluk-knappen.
Bemærk: Displayet skifter til standbytilstand ogdæmpes efter 20 sekunder. Tryk på en vilkårlig knap
for at vække skærmen fra standbytilstand.
9. For avancerede funktioner, se instruktionerne
til menunavigation.

SPECIELLE PRODUKTFUNKTIONER
autoSAT®
CAIREs proprietære autoSAT-funktion gør
det muligt for FreeStyle Comfort at levere enensartet mængde ilt (bolus) pr. vejrtrækning,
op til en maksimal vejrtrækningshastighed.
Efterhånden som brugerens åndedræt stiger underaktiviteter som gang, justerer autoSAT-funktionen
enhedens funktion for at opretholde en indstillet
bolusvolumen.
UltraSense®
Denne sensitive pulsdosisudløser registrerertrykændring i dit åndedræt, og frigør iltindtag
pr. hjerteslag ved indånding.
autoDOSE®
Skulle der forekomme en "ingen vejrtrækning"alarm, vil denne funktion puste ilt ind med
20 vejrtrækninger pr. minut med enhedens aktuelle
flowindstilling, indtil alarmtilstanden er afviklet.

FreeStyle Comfort
Strømforsyninger
FreeStyle Comfort kan forsynes med strøm påtre måder:• Genopladelig batteripakke• Vekselstrøm• Jævnstrøm
Oplysninger om anvendelsen af hver af disse
komponenter findes i de efterfølgende afsnit.
ADVARSEL: BRUG KUN DE STRØMFORSYNINGER, DER ER ANFØRT I
DENNE BRUGERVEJLEDNING. BRUG
AF STRØMFORSYNINGER OG BATTERIPAKKER, DER IKKE ER LEVERET AF
CAIRE, KAN FORÅRSAGE SKADE PÅ
ENHEDEN OG UKORREKT DRIFT.

UDTAGNING AF BATTERIETTræk batteriudløserknappen nedad og væk fra
FreeStyle Comfort. Mens denne knap HOLDES
NEDE, skal du føre batteriet ud ved at trække
det væk fra koncentratoren.
ADVARSEL: ILTKONCENTRATORENSKAL VÆRE SLUKKET OG KOBLET FRA
DEN EKSTERNE STRØMFORSYNING,
INDEN BATTERIPAKKEN TAGES UD.

BATTERIPAKKE

FreeStyle Comfort kan køre på den genopladelige
batteripakke, som følger med enheden. Der er ikke
nogen driftsmæssige begrænsninger af koncentratoren,når den kører på batteripakken. Alle flowhastigheder
kan anvendes, når enheden kører på batteri.

KONTROL AF BATTERIPAKKENSLADETILSTAND
ISÆTNING AF BATTERIET

Ret batteriet ind efter bunden af huset fra midten
af FreeStyle Comfort. Tryk batteriet indad, indtil
du hører et klik, som bekræfter, at det er låst fast.

Hvis du vil kontrollere batteriets ladetilstand, når det
er sat i FreeStyle Comfort, skal du se på batteriikonet
og den anslåede resterende tid på displayet.
80-100%60-80%

40-60%

20-40%

20%

PN MN232-C4 L | Brugermanual DK - 253

FreeStyle ComfortBemærk: Hvis strømstikket er tilsluttet, vises et stikikoni stedet for den resterende ladeprocent.
Hvis du vil kontrollere batteriets ladetilstand,når det ikke er sat i Comfort, skal du lægge
batteriet fladt ned og trykke på knappen TEST.
Batterimålerindikatorerne til højre for knappen
TEST tændes for at angive batteripakkens
ladetilstand (25-100 %).

BATTERIETS DRIFTSTID

Når den eksterne veksel- eller jævnstrømforsyning
frakobles, skifter koncentratoren automatisk til
at køre på batteriet, hvis et opladet batteri er isat.
Nedenstående tabel viser de normale driftstider
for en ny batteripakke. Batteripakkens driftstid
forbliver konstant, uafhængigt af hvor hurtigt
du trækker vejret.
Pulseringsindstilling

Varighed af
enkelt (8-cellet)
batteri

Varighed af
dobbelt (16-cellet)
batteri

1

op til 8 timer

op til 16 timer

2

op til 4 timer

op til 8 timer

3

op til 3 timer

op til 6 timer

4

op til 2,25 timer

op til 4,5 timer

5

op til 2 timer

op til 4 timer

NORMAL BATTERIGENOPLADNINGSTID
DEN FØRSTE OPLADNING

Det nye batteri, som følger med din FreeStyle
Comfort, er kun delvist opladet, når det leveres fra
fabrikken. Du skal lade batteripakken helt op med
enten veksel- eller jævnstrøm, inden du tager det
i brug første gang.

GENERELLE OPLYSNINGER OM
OPLADNING

FreeStyle Comfort genoplader altid batteripakken,
når denne er sat i koncentratoren, og
koncentratoren tilsluttes en ekstern veksel- ellerjævnstrømforsyning (undtagen ombord på et fly).
Batteriet er helt opladet, når det viser 100 %.
Batteripakken kan også oplades med den eksterne
oplader (ekstraudstyr).

254 - DK PN MN232-C4 L | Brugermanual

Den typiske genopladningstid for batteripakken,
hvis den er helt afladet, er 3,5 for et enkelt batteri
og 6,0 timer for et dobbelt batteri.
Bemærk: Batteriet starter muligvis ikke
genopladningen med det samme, hvis
batteripakken er helt afladet.

BATTERILEVETIDFreeStyle Comfort-batterierne er udviklet til atholde til 300 opladnings-/afladningscyklusser.

FreeStyle Comfort
Veksel- og jævnstrømforsyning
FreeStyle Comfort leveres med vekselstrømsog jævnstrømforsyning som standard.
Vekselstrømsforsyningen omfatter følgende:	a. Vekselstrømsblokken, der fungerer mellem
100-240 volt, 50-60 Hz	b. Vekselstrømskablet, der forbinder
strømblokken med stikkontakten.

Jævnstrømforsyningen består af en ledning,
der tilsluttes direkte enheden og jævnstrømskilden.

2. Slut vekselstrømskablet til vekselstrømsblokken.

VEKSELSTRØM

Vekselstrøm bruges derhjemme eller andre steder,
hvor en almindelig stikkontakt står til rådighed.
FreeStyle Comfort er fuldt funktionsdygtig,
når den kører på vekselstrøm. FreeStyle Comfort
genoplader også batteripakken, hver gang den
tilsluttes vekselstrøm (undtagen ombord på fly).
Batteripakken genoplades, uanset om FreeStyle
Comfort er tændt eller slukket.
ADVARSEL: UNDGÅ AT TILSLUTTEEN EKSTERN STRØMFORSYNING, HVISDER IKKE ER SAT EN BATTERIPAKKEI ILTKONCENTRATOREN.
Tilslutning af Comfort til vekselstrøm:
1. Sæt vekselstrømsblokken i strømindgangennederst til højre foran på FreeStyle Comfort.
Sørg for at skubbe den indad, indtil den sidder
forsvarligt fast.

3. Slut den modsatte ende af vekselstrømskablet
til en stikkontakt i væggen eller en anden egnet
stikkontakt.

PN MN232-C4 L | Brugermanual DK - 255

FreeStyle Comfort
4. Tænd for FreeStyle Comfort ved at trykke påtænd/sluk-knappen, og vælg den foreskrevne
flowindstilling ved hjælp af knappen til øgning(+) eller reduktion (-) af flow.

4. Tænd for FreeStyle Comfort ved at trykke påtænd/sluk-knappen, og vælg den foreskrevne
flowindstilling ved hjælp af knappen til øgning(+) eller reduktion (-) af flow.
Bemærk: Hvis batteripakken skal genoplades, harbatteriikonet på displayet en enkelt orange bjælke.

JÆVNSTRØM

Jævnstrøm bruges i tilbehørsstik i biler, bådeeller andre motorkøretøjer. FreeStyle Comfort er
fuldt funktionsdygtig til brug, når den kører påjævnstrøm. FreeStyle Comfort genoplader ogsåbatteripakken, hvis det er nødvendigt, hver gangden tilsluttes jævnstrøm. Batteripakken genoplades,
uanset om enheden er tændt, slukket eller i brug.
ADVARSEL: UNDGÅ AT TILSLUTTE ENEKSTERN STRØMFORSYNING, HVISDER IKKE ER SAT EN BATTERIPAKKE I
ILTKONCENTRATOREN.
1. Start køretøjets motor.
2. Sæt jævnstrømforsyningen i strømindgangennederst til højre foran på FreeStyle Comfort.
Sørg for at skubbe den indad, indtil den sidder
forsvarligt fast.

3. Slut den modsatte ende af jævnstrømskablet
til en bils tilbehørsstik eller til et andet egnet
jævnstrømsudtag.
256 - DK PN MN232-C4 L | Brugermanual

FORSIGTIG: For at forhindre mulig
brand eller overophedning skal du sikre
dig, at 12 V jævnstrøms-stikkontaktener fri for snavs, og at adapterstikket påjævnstrømforsyningen passer korrektog forbliver sikkert på plads.
FORSIGTIG: Jævnstrømforsyningens
metaloverflader kan blive MEGET VARME, når deer i brug på grund af høj strøm. Giv overfladernetilstrækkelig tid til at køle af, før de berøres.
FORSIGTIG: Brug ikke jævnstrømforsyningen medforlængere, splittere eller adaptere. Brug af ikkegodkendt tilbehør kan resultere i overophedningeller beskadigelse.
FORSIGTIG: Tilslutning af enheden til en 12 Vjævnstrøms-stikkontakt kan aflade køretøjets
batteri, hvis motoren ikke kører, selvom enheden
er slukket.
FORSIGTIG: Start ikke et køretøj med starthjælp,mens jævnstrømforsyningen er tilsluttet.

FreeStyle Comfort
Rengøring, pasningog rutinevedligeholdelse
RUTINEVEDLIGEHOLDELSE

Forebyggende vedligeholdelse Enheden er
vedligeholdelsesfri. Filtre inde i enheden er
beregnet til at holde hele enhedens levetid. Hvis
enheden ikke fungerer korrekt, skal enheden
returneres til leverandøren til reparation. Service
af komponenter inde i kabinettet på FreeStyle
Comfort skal udføres af iltleverandøren.
Brugeren af denne enhed bærer ansvaret fornedenstående vedligeholdelsesprocedure.
• Ugentligt – Rengør luftindtagsfiltrene
ADVARSEL: GØR IKKE FORSØG PÅ
AT ÅBNE ELLER FJERNE PLASTIKDÆKSLERNE PÅ ILTKONCENTRATOREN. ENHEDEN MÅ KUN ÅBNES AF
EN KVALIFICERET SERVICETEKNIKER.

LUFTINDTAGSFILTRE

Luften kommer ind i FreeStyle Comfort gennem
luftindtagsfiltre, der er placeret nederst foran på
iltkoncentratoren. Disse filtre forhindrer, at dertrænger støv eller andre store partikler ind i systemet.
Luftindtagsfiltrene skal rengøres mindst én gangom ugen. Din iltleverandør kan eventuelt rådedig til at rengøre de hyppigere afhængigt af
driftsbetingelserne. Udfør trinnene herunder
for at rengøre luftindtagsfiltrene korrekt.
1. Fjern de to filtre fra forsiden af FreeStyle
Comfort ved at trykke den tilsvarende tap opad
og trække den udad.

ADVARSEL: UNDGÅ AT BRUGE
ILTKONCENTRATOREN, UDEN AT
LUFTINDTAGSFILTERET ER ISAT.

PLEJE AF BATTERIPAKKEN

Batteripakken må ikke udsættes for væsker.
Gør ikke forsøg på at adskille eller åbnebatteripakken.
Batteriet må ikke udsættes for åben ild.
Undgå at tabe batteripakken.
Opbevar batteripakken uden for børns rækkevidde.
Hvis der er tegn på beskadigelse af batteripakken,
skal du kontakte din iltleverandør, inden du tager
den i brug.
Batteriet må ikke anbringes i eller i nærheden af en
mikrobølgeovn eller andet husholdningsudstyr.

OPBEVARING AF BATTERIPAKKEN

Det er ikke nødvendigt at aflade batteriet
i iltkoncentratoren helt, før det genoplades.
Det anbefales at oplade batteriet i iltkoncentratoren
efter hver brug.
Batteriet må ikke ligge et sted, hvor det er udsat for
overdreven varme, som f.eks. i direkte sollys eller
i en bil. Hvis det sker, kan det forringe batteriets
ydelse og forkorte dets levetid. Det kan også få
batteriet til at generere varme og røg, bryde i brandeller eksplodere. Batteripakken skal opbevares
i stabile omgivelser med lav luftfugtighed(under 70 % relativ luftfugtighed) og fri forætsende gasser. For at forebygge rustdannelse er
det vigtigt at undgå omgivelser, som kan forårsage
kondensering på grund af eksempelvis hurtigeændringer af den omgivende temperatur.
Ved opbevaring i en måned eller derover anbefales
afladning eller delvis afladning til et niveau mellem
50 % og 75 % (to til tre lysdioder). Tag batteriet ud
af iltkoncentratoren, når det er opladet.

2. Vask begge filtre i en varm opløsning af sæbeog vand.
3. Skyl filtrene grundigt, og fjern overskydende
vand med en blød, fnugfri klud. Sørg for, atfilterene er helt tørre, inden de sættes på plads.
4. Isæt de rene, tørre filtre igen.
PN MN232-C4 L | Brugermanual DK - 257

FreeStyle ComfortRENGØRING

Sluk for enheden, og frakobl veksel- ellerjævnstrømforsyningen, før al rengøring.
Du MÅ IKKE sprøjte direkte på kabinettet.
Brug en fugtig (ikke våd) klud eller svamp.
Spray kluden eller svampen med et mildt
husholdningsrengøringsmiddel for at rengøre
kabinettet og strømforsyningen. Forsæt efteranvisningerne fra producenten af rengøringsmidlet.
ADVARSEL: SLUK FOR ENHEDEN,
OG TRÆK STRØMKABLET UD AFSTIKKONTAKTEN, FØR DU RENGØR
ENHEDEN, FOR AT FORHINDREUTILSIGTET ELEKTRISK STØD OG
FORBRÆNDINGER. FJERNELSE
AF DÆKSLER ELLER SERVICE PÅ
ENHEDEN MÅ KUN UDFØRES AF
LEVERANDØREN AF UDSTYRET ELLEREN KVALIFICERET SERVICETEKNIKER.
ADVARSEL: VÆR OMHYGGELIG MED AT UNDGÅ,AT ILKONCENTRATOREN OG BATTERIPAKKENBLIVER VÅDE, ELLER AT DER KOMMER
VÆSKE IND I ENHEDEN. DETTE KAN MEDFØREFUNKTIONSFEJL ELLER NEDLUKNING SAMTØGET RISIKO FOR ELEKTRISK STØD ELLERFORBRÆNDINGER.
ADVARSEL: UNDGÅ AT BRUGE VÆSKE DIREKTEPÅ ENHEDEN. LISTEN OVER UØNSKEDEKEMISKE STOFFER OMFATTER, MEN ER IKKEBEGRÆNSET TIL, FØLGENDE: ALKOHOLOG ALKOHOLBASEREDE PRODUKTER,
KONCENTREREDE KLORINBASEREDEPRODUKTER (ÆTYLENKLORID) SAMTOLIEBASEREDE PRODUKTER. DISSEMÅ IKKE ANVENDES TIL RENGØRING AF
PLASTIKKABINETTET PÅ KONCENTRATOREN,
DA DE KAN BESKADIGE ENHEDEN.
ADVARSEL: RENGØR KUN KABINETTET,BETJENINGSPANELET OG STRØMKABLET MED
ET MILDT RENGØRINGSMIDDEL PÅ EN FUGTIG(IKKE VÅD) KLUD ELLER SVAMP, OG AFTØR
ALLE OVERFLADER. UNDGÅ, AT DER KOMMER
VÆSKE IND I KONCENTRATOREN. VÆR SÆRLIG
OPMÆRKSOM PÅ ILTUDTAGET TIL KANYLETILSLUTNINGEN FOR AT SIKRE, AT DEN ER FRI FOR
STØV, VAND OG PARTIKLER.

258 - DK PN MN232-C4 L | Brugermanual

ADVARSEL: BRUG EN FUGTIG (IKKE VÅD)KLUD ELLER SVAMP TIL AT RENGØREBATTERIPAKKEN. SPRAY FØRST KLUDEN ELLER
SVAMPEN MED ET MILDT RENGØRINGSMIDDEL,
OG RENGØR BATTERIPAKKENS HUS.
ADVARSEL: UNDGÅ AT RENGØRE VED HJÆLP
AF ET AUTOMATISK VASKESYSTEM.

UDSKIFTNING AF KANYLEN

Følg altid kanyleproducentens vejledning forkorrekt brug. Udskift næsekanylen eller
iltslangerne hver 3. til 6. måned (hver 3. måned
ved daglig brug) i henhold til kanyleproducentens
eller iltleverandørens anbefaling. Din læge eller
iltleverandør informerer dig om rengøring,desinfektion og udskiftning.
Du kan få yderligere udskiftningsdele hos diniltleverandør.

FreeStyle Comfort
Alarmtilstande
Prioritetnr.

Alarmbeskrivelse

1

Systemfejl,
batteri høj
temperatur

Display

Lydsignal

Flere detaljer/anbefalet handling

1 bip hvert
20. sek.

Enheden udløser en alarm, når batteriets
temperatur nærmer sig tærsklen for sikkerhedsnedlukning.
Enheden lukker ned, hvis nedlukningstærsklen overskrides. Hvis en ekstern
strømkilde er tilsluttet, vises alarmskærmbilledet, og der udsendes en lydalarm, som
fortsætter i 2 min.
Test med batteri ved stuetemperatur. Kontakt leverandøren, hvis problemet vedvarer.

2

Systemfejl,
høj temperatur

1 bip hvert
20. sek.

Enheden udløser en alarm, når den
indvendige temperatur nærmer sig
nedlukningstærsklen.
Hvis nedlukningstærsklen overskrides,
lukker enheden ned, og alarmskærmbilledet vises, mens der udsendes en lydalarm,som fortsætter i 2 min.
Sørg for, at enhedens ventilationssprækker
ikke blokeres af noget. Kontakt leverandøren, hvis problemet vedvarer.

3

Systemfejl,
højt tryk (højtsystemtrykeller kompressorfejl)

1 bip hvert
20. sek.

4

Systemfejl,
flowhastighed(ventilsvigt iiltkoncentrator, blokering
af gasvej)

1 bip hvert
30. sek.

Kontakt leverandøren, hvis problemetvedvarer.

5

Nedlukningpga. lavt
batteriniveau

1 bip hvert
30. sek.

Alarmen fortsætter, indtil batteriet løber tør
for strøm, og enheden lukkes ned.

Hvis nedlukningstærsklen overskrides,
lukker enheden ned, og alarmskærmbilledet vises, mens der udsendes en lydalarm,som fortsætter i 2 min.
Kontakt leverandøren, hvis problemetvedvarer.

Genoplad batteriet.

PN MN232-C4 L | Brugermanual DK - 259

FreeStyle Comfort
Prioritetnr.

Alarmbeskrivelse

6

Ingen
vejrtrækning
registreret

Display

Lydsignal

Flere oplysninger

1 bip hvert
10. min.

Ingen vejrtrækning detektereti 15 sekunder.
Aktiveres efter første vejrtrækning, eller når
opvarmningsskærmbilledet er slukket, alt
efter hvad der kommer først.
Kontroller kanylen for knæk eller lækager.
Forøg følsomhedsindstillingen. Kontakt
leverandøren, hvis problemet vedvarer.

7

Opvarmning

Intet

Enheden starter med at dosere automatisk,
indtil der detekteres en vejrtrækning, eller
opvarmningen slutter, alt efter hvad derkommer først.

8

Lav
koncentration

1 bip hvert
10. mini op til
30 min.

Kontakt leverandøren, hvis problemetvedvarer.

9

Advarsel om
lavt batteriniveau

1 bip,
ingengentagelse

Genoplad batteriet.

10

Respirationsfrekvens
overskredet

1 bip,
ingengentagelse

Respirationsfrekvensen overskrider
40 vejrtrækninger pr. minut.

11

Systemfejl,
ventilatorsvigt

Intet

Kontakt leverandøren, hvis problemetvedvarer.

12

Systemfejl,
lavt tryk

Intet

Kontakt leverandøren, hvis problemetvedvarer.

13

Systemfejl,
lav temp.

Intet

Lad enheden varme op til stuetemperatur.
Kontakt leverandøren, hvis problemetvedvarer.

260 - DK PN MN232-C4 L | Brugermanual

Hvis der udløses dosis utilsigtet, skal
følsomhedsindstillingen forøges. Kontakt
leverandøren, hvis problemet vedvarer.

FreeStyle Comfort
Prioritetnr.

Alarmbeskrivelse

Info

Opstart

Display

Lydsignal

Flere oplysninger

1 infobip

Dette skærmbillede vises under opstarten,og du bør høre et bip, som bekræfter,
at brummeren fungerer.
Opstartsskærmbilledet viser den aktuelle
firmwarerevision i nederste højre hjørne.

InfoSkærmbillede fornormal drift
i autoSAT-tilstand (ingenjævnstrøm
tilsluttet)

Intet

Info

Skærmbillede
for normal
drift med
autoSAT-tilstand deaktiveret (ingenjævnstrøm
tilsluttet)

Intet

Info

Indstillingændret1 infobip

Info

Tilslut ekstern
strømkilde
i fra-tilstand

Intet

Info

Tilslut ekstern
strømkildei kørselstilstand

Intet

Info

Frakobleksternstrømkilde
i fra-tilstand

1 infobip

CAIRE-skærmbilledet vises i 2 sekunder,
efterfulgt af batteri- og stikstatus i 5 sek.,før displayet slukkes.

CAIRE-skærmbilledet vises i 2 sekunder,
hvorefter displayet slukkes.

PN MN232-C4 L | Brugermanual DK - 261

FreeStyle Comfort
Prioritetnr.

Alarmbeskrivelse

DisplayInfo

Frakobleksternstrømkildei kørselstilstand (medbatteri isat)

1 infobipInfo

I standbytilstand – når
der trykkes
på op- eller
nedknappen,
viser displayet batteri- og
ladetilstand

Intet

Info

Flytilstandmed
batteristrøm

1 langt bip

Info

Flytilstandmedtilslutning til
stikkontakt

1 langt bip

262 - DK PN MN232-C4 L | Brugermanual

Lydsignal

Flere oplysningerSkærmbilledet vises i 5 sek.,
hvorefter displayet slukkes.

FreeStyle Comfort
Instruktioner vedr. menunavigation
Du kan åbne menuen i tilstanden Standby eller
kørselstilstand. Hvis der udløses en alarmtilstandi kørselstilstand, afsluttes menuen automatisk.
Hvis der ikke trykkes på nogen knapper i
20 sekunder, afsluttes menuen automatisk.
I standby skal du muligvis trykke to gange påknappen Menu for at vække enheden og derefteråbne menuen.
Der er fire menuskærmbilleder:

Flytilstand1. Statusskærm: Viser firmwarens revisionsnummer,
driftstid, batteriafladningscyklusser
2. Flytilstand (TIL/FRA)
3. Følsomhed: justerbar (1-5)
4. autoSAT (TIL/FRA)

FølsomhedEksempel på statusskærmbillede

BETJENING AF INDSTILLINGSMENUEN

autoSAT

Indstillingsmenuen åbnes ved at trykke på knappen
Menu/Vælg

i tilstanden Kør eller Standby.

Løb igennem menuerne ved at trykke på.

StatussidePN MN232-C4 L | Brugermanual DK - 263

FreeStyle Comfort
STATUSSIDE
Statussiden på Freestyle Comfort viser
firmwarerevision, driftstimer og antal
battericyklusser på det tilsluttede batteri.

Annuller flytilstand ved at trykke påmens ikonet ANNULLER er synligt.

,

Flytilstand aktiveret bekræftet

Statusside

Driftstimer vises i hele timer.
Batteriafladninger er antallet af cyklusser, hvor det
tilsluttede batteri er blevet afladet og genopladet.

FLYTILSTAND
FreeStyle Comfort leveres med flytilstanddeaktiveret.

Deaktiver flytilstand ved at gentage ovenstående,
mens du skifter til FRA.
Tryk på
. Når skærmbilledet viser symbolet"Næste", overgår menusiden til det næste skærmvalg.
Flytilstand er beregnet til brug i kørselstilstand.
Flytilstand deaktiveres automatisk, hvis enheden
efterlades i standby på batteristrøm.
Bemærk: Se afsnittet Aktiver flytilstand for at fåyderligere oplysninger om flytilstand.

FØLSOMHEDSTILSTAND

Hovedskærmbillede for flytilstand

Aktiver tilstanden ved at trykke på,mens ikonet SKIFT er synligt. Værdien skifter
til det blå ord "TÆNDT".

FreeStyle Comfort leveres med en indstilling forstandard følsomhedstilstand på 2 fra fabrikkens
side og burde ikke kræve justering under normale
forhold. Følsomhedsindstillingen kan justeresi værdiintervallet 1-5, hvor 1 er det mindst
følsomme, og 5 er det mest følsomme.

Skærmbilledet Følsomhedstilstand ved indstillingen 2

Flytilstand aktiveret valgtBekræft flytilstand ved at trykke på,mens ikonet GEM er synligt. Værdien skifter
tilbage til ordet "TÆND".

264 - DK PN MN232-C4 L | BrugermanualFreeStyle Comfort
Juster følsomhedstilstanden ved at trykke på,mens ikonet SKIFT er synligt. Værdien skifter til
det næste blå tal.

Følsomhedstilstand indstilling 3 valgtBekræft indstilling 3 i følsomhedstilstand ved
at trykke på, mens ikonet GEM er synligt.
Værdien skifter tilbage til "3".

TILSTANDEN AUTOSATFreeStyle Comfort leveres med tilstanden autoSAT
aktiveret.

Hovedskærmbillede for autoSAT

Deaktiver tilstanden ved at trykke på,mens ikonet SKIFT er synligt. Værdien skifter til
det blå ord "SLUKKET".

Annuller Følsomhedstilstanden ved at trykke på, mens ikonet ANNULLER er synligt.

autoSAT-tilstand deaktiveret er valgtBekræft tilstanden autoSAT ved at trykke på
Følsomhedstilstand indstilling 3 bekræftet

, mens ikonet GEM er synligt. Værdien skifter
tilbage til ordet "SLUK".
Annuller tilstanden autoSAT ved at trykke påmens ikonet ANNULLER er synligt.

,

Følsomhedstilstand indstilling 5 (maks.) bekræftet

Tryk på
. Når skærmbilledet viser symbolet"Næste", overgår menusiden til det næste
skærmvalg.

Tilstanden autoSAT deaktiveret bekræftet

Aktiver flytilstand ved at gentage ovenstående,
mens du skifter til TIL.
Tryk på
. Når skærmbilledet viser symbolet"Næste", overgår menusiden til det næste
skærmvalg.

PN MN232-C4 L | Brugermanual DK - 265

FreeStyle Comfort
Fejlfinding
ProblemMulig årsag

Løsning

FreeStyle Comfort viser
alarm og udsenderperiodiske bip.

Se tabellen med
alarmtilstande.

Se tabellen med alarmtilstande.

Iltkoncentratorentændes ikke.

Batteriet er afladet.

Tilslut enheden til en jævnstrøm- eller
vekselstrømforsyning.

Den er ikke sluttet tilekstern strømforsyning.
Generel fejlfunktion.

Kontrollér, at de eksterne tilslutninger er sikre.
Udfør en komplet sluk/tænd-cyklus ved at tagebatteriet ud og frakoble en eventuel ekstern vekseleller jævnstrømskilde. Sæt derefter batteriet i igen,hvorefter sluk/tænd-cyklussen er fuldført. Tilslut
eventuelt den eksterne strømkilde igen.
Kontakt udstyrsleverandøren, og skift om
nødvendigt til en anden iltkilde.

Ved alle andre problemer.

Udfør en komplet sluk/tænd-cyklus ved at tagebatteriet ud og frakoble en eventuel ekstern vekseleller jævnstrømskilde. Sæt derefter batteriet i igen,hvorefter sluk/tænd-cyklussen er fuldført. Tilslut
eventuelt den eksterne strømkilde igen.
Skift til en anden iltkilde, og kontakt
udstyrsleverandøren.

Bemærk: Nogle alarmer får enheden til at lukke ned.
Bemærk: Under en total strømafbrydelse nulstilles alarmsystemet (rydder alle tidligere alarmer),og der vises kun en alarm, hvis tilstanden stadig er aktuel, når strømmen genoprettes.

266 - DK PN MN232-C4 L | BrugermanualFreeStyle Comfort
Oplysninger om rejser

1. Tænd KUN POC med batteristrøm, når der ikke
er tilsluttet en ekstern strømkilde.

FreeStyle Comfort imødekommer FAA's
retningslinjer vedr. brug på ruteflyvninger. Forudfor enhver rejse (med fly, tog, skib osv.) skal du:• sørge for, at koncentratoren er i god stand og
driftsklar.
• kontakte rejseselskabet mindst 48 timer før
afrejsen for at informere dem om, at du rejser
med en FreeStyle Comfort.
Det anbefales, at du altid medbringer ekstra
batterier på rejser. I forbindelse med flyrejser skal
du sørge for at have tilstrækkeligt batteri til hele
forløbet (check-in før flyvning, sikkerhedstjekosv.) samt mellemlandinger. De fleste flyselskaberkræver, at du har et tilstrækkeligt antal fuldt
opladede batteripakker til drift af enheden i mindst
150 % af turens forventede maksimale varighed(flyvetider, al tid på jorden, før og efter flyvningen,under forbindelser samt til uventede forsinkelser).
Bemærk: På flyrejser anbefales det at medbringede ekstra batterier i en sikker bæretaske.

FAA-REGULERING

For at forhindre kortslutninger skal de ekstra
batterier beskyttes individuelt ved at blive anbragt
i den oprindelige emballage eller ved anden
terminalisolering, f.eks. ved at sætte tape hen overudsatte terminaler eller ved at anbringe hvert batteri
i en separat plastikpose eller et beskyttelsesetui,og de må kun transporteres i håndbagagen.

2. Aktiver flytilstand ved brug af én af følgende
metoder:a. Følg instruktionerne for at aktivere flytilstandi afsnittet Instruktioner vedr. menunavigation.

b. Tryk på op- og nedknapperne, og hold dem
nede i 5 sekunder, indtil du hører et langt bip,og symbolet for flytilstand vises.

AKTIVÉR FLYTILSTAND
Batteristrøm
POC kan bruges ombord på fly i henhold til
de normale betjeningsprocedurer, KUN med
batteristrøm, hvis der ikke er tilsluttet en ekstern
strømkilde.
Eksternt strømHvis der er en tilgængelig stikkontakt ombord på etfly, og den er placeret, så den er praktisk at brugesom POC-strømkilde, kan den fungere som reserve
for POC-batterierne. I dette tilfælde skal følgende
anbefalede procedure for skift fra batteri til strøm
fra flyet følges:

3. Når flysymbolet vises, skal du slutte POC til
stikkontakten.

Bemærk: Ombord på fly er det ikke muligt at opladebatterierne i flytilstand.

PN MN232-C4 L | Brugermanual DK - 267

FreeStyle Comfort
Tilbehør

PLACERING AF SYSTEMETI BÆRETASKEN

Anvend kun tilbehør på listen, som er leveret af
CAIRE via din iltleverandør, af hensyn til korrekt
ydeevne og sikkerhed. Brug af tilbehør, der ikke
er angivet nedenfor, kan påvirke ydeevnen og/eller
sikkerheden af koncentratoren.

1. Anbring systemet i bunden af tasken, og sørg forat placere filtrene og batteriåbningerne korrekt.

Standardtilbehør til FreeStyle ComfortBæretaske

MI396-1RygsækMI459-1

Skulderrem

MI397-1

Vekselstrømsforsyning(alle ledninger medfølger)

PW036-1SJævnstrømskabel

CD041-1

Eksternt grovpartikelfilter (højre)FI226-1

Eksternt grovpartikelfilter (venstre)

FI227-1

2. Luk lynlåsen rundt om bunden af tasken.

Valgfrit ekstraudstyr
Ekstra 8-cellet batteripakke

BT034-1S

16-cellet batteripakke

BT035-1S

Ekstern oplader til batteripakker

BT036-1

Tilbehørstaske

MI372-2Bemærk: Der kan være flere valgmuligheder med
henblik på landespecifikke strømkabler, som anført
ovenfor. Kontakt CAIRE eller din iltleverandør, hvis
du har brug for flere valgmuligheder i forbindelse med
en bestilling.

Instruktioner til bæretasken
Følg nedenstående instruktioner for at placereog fastgøre dit FreeStyle Comfort-systemi bæretasken.

268 - DK PN MN232-C4 L | Brugermanual

MONTERING AF SKULDERREM
1. Skulderremmen kan fastgøres direktepå enheden eller på bæretasken.
2. Sæt remmens clips fast på enhedens eller
taskens ring.

FreeStyle Comfort
Bortskaffelse og genbrug
Aflever altid FreeStyle Comfort med allekomponenter til din leverandør med henblik
på bortskaffelse. Du kan også kontakte den
lokale myndighed angående instruktioner om
korrekt bortskaffelse af batteriet. Enheden kan
genbruges af en anden person, men skal returneres
til udstyrsleverandøren til oparbejdning.

WEEE og RoHS
Dette symbol har til formål at minde ejere
af udstyr om at returnere det til et genvindingsanlæg, når det er udtjent, i henhold til direktivet om
affald af elektrisk og elektronisk udstyr (WEEE).
Vores produkter overholder begrænsningerne
i RoHS-direktivet. De indeholder ikke mere
end spor af bly eller andre farlige stoffer.

Materialer, der kommer i direkte
berøring med brugeren• Koncentratorens kabinet: PC/ABS• Koncentratorens betjeningspanel:
Polyester EBG7 eller tilsvarende• Iltudtag, dyse: Aluminium• Enhedens mærkat: Lexan• Ledningsstik: Polycarbonat/vinylklorid• Netledning: PVC, metal• Strømforsyning: Lexan 940 (polycarbonat)• Batteripakke: PC/ABS• Batteripakke, strømforsyningernes mærkater:
Polyesterfilm• Koncentratorens bæretaske: Cordura, neopren,nylonfor• Rem: Polypropylen, acetyl, vinyl, Cordura,
neoprenFORSIGTIG: Kontakt leverandøren
af udstyret eller din lokale offentlige
institution for at høre om kravene
til bortskaffelse.
FORSIGTIG: Bortskaf altid molekylsigten
i overensstemmelse med alle føderale,statslige og lokale miljøkontrolbestemmelser.

PN MN232-C4 L | Brugermanual DK - 269

FreeStyle Comfort
EMC-test
Vejledning og producentens erklæring – elektromagnetiske emissioner
FreeStyle Comfort er beregnet til brug i det elektromagnetiske miljø, der er angivet nedenfor. Kunden eller
brugeren af FreeStyle Comfort er ansvarlig for at sikre, at det anvendes i et sådant miljø.
Emissionstest
RF-emissioner
CISPR 11
RF-emissioner
CISPR 11
Harmoniske emissioner
IEC 61000-3-2Spændingsudsving/flickeremissioner
IEC 61000-3-3Overensstemmelse

Elektromagnetisk miljø – vejledning

Gruppe 1

FreeStyle Comfort anvender kun RF-energi til interne funktioner.
Derfor er RF-emissionerne meget lave og vil sandsynligvis ikkeforårsage interferens i elektronisk udstyr i nærheden.

Klasse B
Klasse A
Overholder

270 - DK PN MN232-C4 L | Brugermanual

FreeStyle Comfort er egnet til brug i alle bygninger, såvel
privatboliger som bygninger, der er direkte forbundet til det
offentlige lavspændingsnet, som forsyner beboelsesejendomme.

FreeStyle Comfort
Vejledning og fabrikantens erklæring – elektromagnetisk immunitet

FreeStyle Comfort er beregnet til brug i det elektromagnetiske miljø, der er angivet nedenfor. Kunden eller brugeren
af FreeStyle Comfort er ansvarlig for at sikre, at det anvendes i et sådant miljø.
Immunitetstest
IEC 60601-testniveau Overensstemmelses- Elektromagnetisk miljø – vejledning
niveauGulvene bør være af træ, beton eller
Elektromagnetisk±8 kV-kontakt±8 kV-kontaktkeramiske fliser. Hvis gulvene er bemiljø – vejledningklædt med syntetisk materiale, skal den±15 kV luftudladning±15 kV luftudladning
IEC 61000-4-2
relative fugtighed være mindst 30 %.
±2 kV for±2 kV forstrømledninger
Elektrisk hurtig
Strømkvaliteten skal være den samme
strømledninger
transient/burst
som i typiske hospitals- og hjemmeplejemiljøer.
IEC 61000-4-4±1 kV for input/outputIkke relevant
ledninger±1 kV linje(r) til linje(r)Strømstød
IEC 61000-4-5Spændingsfald,korte afbrydelser
og spændingsvariationer på
strømforsyningens
indgangsledninger
IEC 61000-4-11±2 kV normal tilstand på ±1 kV linje(r) til linje(r)vekselstrømsledninger±1 kV differensstrøm på
vekselstrømsledninger

Strømkvaliteten skal være den samme
som i typiske hospitals- og hjemmeplejemiljøer.

±2 kV normal tilstand på
udendørs I/O-ledninger
0 % UT i 0,5 cyklus (0°, 0 % UT i 0,5 cyklus (0°,45°, 90°, 135°, 180°,
45°, 90°, 135°, 180°,
225°, 270°, 315°)
225°, 270°, 315°)

Strømkvaliteten skal være densamme som i typiske hospitals- og
hjemmeplejemiljøer. Hvis brugeren af
0 % UT i 1 cyklus (0°)
0 % UT i 1 cyklus (0°)
FreeStyle Comfort kræver uafbrudt
drift under strømafbrydelser, anbefales
70 % UT (30 % fald i UT) 70 % UT (30 % fald i UT) det, at FreeStyle Comfort forsynes med
i 25/30 cyklusser (0°)
i 25/30 cyklusser (0°)
strøm fra en nødstrømsforsyning eller
et batteri.
0 % UT i 250/300
cyklusser (0°)

0 % UT i 250/300
cyklusser (0°)

Driftsfrekvens
Netfrekvensmagnetfelterne skal være(50/60 Hz)
på et niveau svarende til niveauet på
3 A/m
3 A/m
magnetfelt
en typisk placering i et hospitals- eller
IEC 61000-4-8
hjemmeplejemiljø.
BEMÆRK UT er vekselstrøms-netspændingen før anvendelse af testniveauet.

BEMÆRK I nogle tilfælde kan en betydelig elektrostatisk udladning medføre, at FreeStyle Comfort nulstilles og kræver
en komplet sluk/tænd-cyklus, før normal drift kan genoptages. Se afsnittet Fejlfinding for at få flere oplysninger.

PN MN232-C4 L | Brugermanual DK - 271

FreeStyle Comfort
Vejledning og fabrikantens erklæring – elektromagnetisk immunitet

FreeStyle Comfort er beregnet til brug i det elektromagnetiske miljø, der er angivet nedenfor. Kunden eller brugeren
af FreeStyle Comfort er ansvarlig for at sikre, at det anvendes i et sådant miljø.
Immunitetstest IEC 60601-testniveau
OverensstemmelsesElektromagnetisk miljø – vejledning
niveau10 VrmsBærbart og mobilt RF-kommunikationsudstyr
10 Vrms
Ledet RFmå ikke anvendes tættere på nogen del af
150 kHz til 80 MHz
IEC 61000-4-6
FreeStyle Comfort, herunder kabler, end
den anbefalede afstand beregnet ud fra denligning, som gælder for senderfrekvensen.
10 V/m
10 V/mUdstrålet RF
IEC 61000-4-3

80 MHz til 2700 MHz

Anbefalet sikkerhedsafstandd = 1.2 P
d = 1.2 P = 80 MHz til 800 MHz
d = 1.2 P = 800 MHz til 2700 MHz
hvor P er senderens maksimale
udgangseffekt i watt (W) i henhold til
producenten af senderen, og d er den
anbefalede afstand i meter (m).
Feltstyrker fra faste RF-sendere,som fastsat af en elektromagnetisk
undersøgelse, a skal være mindre end
overensstemmelsesniveauet i hvert
frekvensområde. b
Interferens kan forekomme i nærheden af​
udstyr mærket med følgende symbol:

BEMÆRK 1 Ved 80 MHz og 800 MHz gælder det højere frekvensområde.
BEMÆRK 2 Disse retningslinjer gælder muligvis ikke i alle situationer. Elektromagnetisk interferens påvirkes af
absorption og refleksion fra bygninger, genstande og mennesker.
aFeltstyrker fra faste sendere, f.eks. basestationer til radiotelefoner (mobile/trådløse) og landmobile radioer, amatørradio, AM- og FM-radioudsendelser og tv-udsendelser kan ikke forudsiges teoretisk med nøjagtighed. Det bør
overvejes at foretage en elektromagnetisk undersøgelse for at vurdere det elektromagnetiske miljø, som skyldes
faste RF-sendere. Hvis den målte feltstyrke på det sted, hvor FreeStyle Comfort anvendes, overstiger ovenstående
gældende RF-niveau, bør det kontrolleres, at FreeStyle Comfort fungerer normalt. Hvis der konstateres unormal
ydeevne, kan yderligere forholdsregler være nødvendige, såsom at vende eller flytte FreeStyle Comfort.
bOver frekvensområdet 150 kHz til 80 MHz skal feltstyrken være mindre end 10 V/m.

272 - DK PN MN232-C4 L | BrugermanualFreeStyle Comfort
Anbefalet separationsafstand mellem bærbart og mobilt RF-kommunikationsudstyr
og FreeStyle Comfort

FreeStyle Comfort er beregnet til brug i et elektromagnetisk miljø, hvori udstrålede RF-forstyrrelser er under kontrol.
Kunden eller brugeren af FreeStyle Comfort kan hjælpe til med at forhindre elektromagnetisk interferens ved atbevare en minimumsafstand mellem bærbart og mobilt RF-kommunikationsudstyr (sendere) og FreeStyle Comfort,
som anbefalet nedenfor, i henhold til kommunikationsudstyrets maksimale udgangseffekt.
Senderens maksimale
Sikkerhedsafstand i henhold til senderens frekvens
udgangseffektm
150 kHz til 80 MHz

80 MHz til 800 MHz

800 MHz til 2,5 GHzW= 1.2 P= 1.2 Pd = 1.2 P
d
d
0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
For sendere med en maksimal udgangseffekt, der ikke er angivet ovenfor, kan den anbefalede sikkerhedsafstand(d) i meter (m) beregnes ved hjælp af den ligning, der gælder for den pågældende senders frekvens, hvor P er den
maksimale udgangseffekt for senderen i watt (W) i henhold til producenten af senderen.
BEMÆRK 1 Ved 80 MHz og 800 MHz gælder sikkerhedsafstanden for det højere frekvensområde.
BEMÆRK 2 Disse retningslinjer gælder muligvis ikke i alle situationer. Elektromagnetisk interferens påvirkes
af absorption og refleksion fra bygninger, genstande og mennesker.

Klassifikation

Beskyttelsestype mod elektrisk stød:
Klasse II	Beskyttelse mod elektrisk stød er			opnået ved dobbelt isolering.

• CISPR 11/EN 55011 Klasse B gruppe 1,
	"Industrielt, videnskabeligt og medicinsk (ISM)	udstyr".
• FCC-del 15, underdel B – klasse B	hændelige strålingskilder.

Grad af beskyttelse mod elektrisk stød:	Type BF Udstyr, der leverer en bestemt
		grad af beskyttelse mod elektrisk
		stød med hensyn til		1) tilladt lækstrøm;		2) pålidelighed af jordforbindelsen		(hvis til stede).
		Ikke beregnet til direkte hjerteanvendelse.

IP22 – Beskyttelse mod indtrængning af fingre
og lignende genstande, der har en diameterpå over 12,5 mm. Vertikale dråber må ikke
have skadelige virkninger, når udstyret haren hældningsvinkel op til 15° i forhold til
vertikallinjen.

Uafhængig prøvning iht. Standard for
Elektromedicinsk udstyr:	Testet til at være i overensstemmelse med• IEC 60601-1 udgave 3.1: Elektromedicinsk
udstyr – Del 1 Generelle krav til sikkerhed.
• CAN/CSA C22.2 No. 60601-1-8 udgave 2.1,
Medicinsk elektrisk udstyr – Del 1: Generelle
krav til sikkerhed.

Tilladt metode til rengøring og infektionskontrol:
Se afsnittet "Rengøring, pasning og korrekt vedligeholdelse" i denne FreeStyle-brugermanual.
Sikkerhedsgrad ved anvendelse undertilstedeværelse af brændbare anæstesigasser:
	Udstyret egner sig ikke til en sådan anvendelse.
Funktionsmetode: Kontinuerlig anvendelse.

Beskyttelse mod potentiel elektromagnetisk eller
anden interferens mellem udstyret og andre enheder.
	• Testet til at være i overensstemmelse med
EN 60601-1-2 /IEC 60601-1-2: 4. udgave.
	• Testet til at være i overensstemmelse med
RTCA/DO160 afsnit 21 kategori M.
PN MN232-C4 L | Brugermanual DK - 273

FreeStyle Comfort
Brukerkontroller og indikatorer for systemstatus
ISO 7000
Brukerhåndbok, instruksjoner for
bruk. Reg.nr. 1641
Landet og produksjonsdatoen.
«CC» identifiserer landskoden på
to bokstaver for produksjonslandet.
Produksjonsdatoen er i formatetÅÅÅÅ-MM-DD. Reg.nr. 6049

ISO 7010: Grafiske symboler – Sikkerhetsfarger
og sikkerhetsskilter – Registrerte sikkerhetsskilt
Instruksjonsmanualen må leses.
Reg.nr M002
Holdes unna åpen flamme, brann,
gnister. Åpne tennkilder og røyking
forbudt. Reg.nr. P003

Må holdes unna regn – oppbevares
tørt. Reg.nr. 0626

Røyking forbudt nær enheten eller
mens enheten brukes. Reg. nr.P002

Stablebegrensning etter antall, der«n» indikerer maksimalt antall
enheter som er tillatt. Reg.nr. 2403

Advarsel. Reg.nr. W001
IEC 60417
Anvendt del type BF (grad av
beskyttelse mot elektrisk støt).
Reg.nr. 5333

Katalognummer. Reg.nr. 2493
Serienummer. Reg.nr. 2498
Lagrings- og driftstemperaturområde.
Reg.nr. 0632
Luftfuktighetsområde ved lagringReg.nr. 2620

Klasse II utstyr, dobbeltisolert.
Reg. nr.5172
EU-forordning 2017/745
Autorisert representant i EU

Produsentens navn og adresse.
Reg.nr. 3082

Hvis produktets unikeenhetsidentifikatormerke (UDI) har
CE####-symbolet, oppfyller enheten
kravene angående medisinsk utstyr.
CE####-symbolet angir nummer for
teknisk kontrollorgan.
Ekstra symboler

Denne siden opp. Reg.nr. 0623

Må holdes unna brennbare materialer,olje og smørefett.

Atmosfærisk trykkbegrensning.
Reg.nr. 2621

Må ikke demonteres.

Glass – håndteres med forsiktighet.
Reg.nr. 0621

Øk oksygentilførsel.

Inneholder farlige stoffer.
Reg.nr. 3723

Reduser oksygentilførsel.

Importør. Reg.nr. 3725

Menyknapp.
O2-uttakStrømbryter. Reg.nr. 5009
RTCA/DO160 avsnitt
21 Samsvarskategori M

274 - NO PN MN232-C4 L | Bruksanvisning

Produsenten av denne POC har bestemt
at denne enheten er i samsvar med allegjeldende FAA-godkjenningskriterierog bruk om bord på fly.

FreeStyle Comfort
Produsenten av denne POC har bestemt
at denne enheten er i samsvar med allegjeldende FAA-godkjenningskriterierog bruk om bord på fly.

Rådsdirektiv 2012/19/EF
WEEE
Dette symbolet skal påminne eierne
av utstyret om at utstyret skal
leveres til et gjenvinningsanlegg når
brukstiden er omme, i henhold til
direktiv om avfall fra elektrisk og
elektronisk utstyr (WEEE-direktivet).
Produktene våre skal være i samsvar
med begrensning av farlige stoffer(RoHS-direktivet). De vil ikke
inneholde mer enn spormengder av
bly eller andre farlige materialer.

Symbol for gjenvinning
Autorisert representant i Sveits(ADD)
Hvis enheten bærer UKCA-merket
som vist med UKCA#### som
indikerer det varslede organets
nummer, er denne enheten i samsvar
med UKCA-forskriftene.
21 CFR 801.15
Amerikansk føderal lovgivningbegrenser denne enheten til salg fra
en lege eller foreskrevet av en lege.

IEC 60601-1-8
Lavprioritert teknisk alarm
ISO 15223-1

ETL-sertifisering
Sertifisert for både det amerikanske
og det kanadiske markedet, i henhold
til gjeldende amerikanske og
kanadiske standarder.
Inneholder FCC ID: WAP2001
FCC-melding
Denne enheten kan inneholde
CYBLE-022001-00, inkludert antennen
2450AT18B100 fra Johnson Technology, somsamsvarer med par. 15 i FCC-reglene. Enheten
oppfyller kravene til godkjenning av modulære
sendere, som beskrevet i FCCs offentlige
melding DA00-1407. Drift av senderen er
underlagt to følgende vilkår: (1) denne enheten
kan ikke forårsake skadelig interferens, og(2) denne enheten må tåle enhver mottatt
interferens, inkludert interferens som kanforårsake uønsket drift. Dette produktet kanvære dekket av ett eller flere amerikanske ellerinternasjonale patenter. Gå til vår hjemmesidenedenfor for en liste med gjeldende patenter.
Pat: www.caireinc.com/corporate/patents

Medisinsk enhet. Reg.nr. 5.7.7
Unik enhetsidentifikator nr. 5.7.10
A S/NZS 4417.1, AS/NZS 4417.2
ACMA Regulatory Complianc
Mark – indikerer samsvar med
ACMA-regelverket (ADD)Trådløs teknologi
Kontakttyper

SSP, iAP2, GATT

Frekvens

2402 til 2480 MHz

Maks RF-utgangseffekt+4 dBmFunksjonsområde

10 m (klasse 2)

Det er anbefalt at maskinen er på en maksimal
avstand på 1,1 cm (0,43”) fra kroppen under drift.
Ikke aktuelt for masker, luftrør eller tilbehør.

Grafiske symboler for bruk på utstyr IEC 60417
Likestrøm reg. nr. 5031
Vekselstrøm reg. nr. 5032
IEC 60601-1
Dråpetett

PN MN232-C4 L | Bruksanvisning NO - 275

FreeStyle Comfort
Utpakking av FreeStyle Comfort
Kontroller at alle komponentene
i listen medfølger i pakningen.
Dersom noen elementer
mangler, ta umiddelbart kontakt
med oksygenleverandøren din.

Standard tilbehør• Bærbar oksygenkonsentrator• Bæreveske• Skulderstropp• Batteripakke med 8 celler• Vekselstrømforsyning• Strømledning til vekselstrøm• Likestrømtilførsel
• Bruksanvisning

Valgfritt tilbehør• Ytterligere batteripakke med
8 celler• Batteripakke med 16 celler• Bordlader• Tilbehørsveske

Gjør deg kjent med FreeStyle Comfort
Gjør deg kjent med de viktigste funksjonene og brukergrensesnittet til FreeStyle Comfort.

Sett forfra
Stroppfeste

Brukergrensesnitt

Oksygenutløp

Luftutslipp

Luftinntak/filtre
Luftinntak/filtre

Knapp forstrømporttaket
Batteripakke
Utløserknapp forbatteripakke
276 - NO PN MN232-C4 L | Bruksanvisning

FreeStyle Comfort
Sett ovenfra
SkjermØk oksygentilførsel

På/av-knapp

Meny-knapp

Reduseroksygentilførsel

Skjerm på batteri
Innstillingen for
strømning

Batterivisning

autoSAT-indikator
Gjenværendebatteriprosent

Alarmvisning
Bolus-leveringsindikator

Skjerm på veksel-/likestrøm
Innstillingen for
strømning

Visningsområde
for alarm

Batterivisning

Ladeindikator

PN MN232-C4 L | Bruksanvisning NO - 277

FreeStyle Comfort
Hva er oksygenkonsentratoren
Luften vi puster inn inneholder omtrent 21 %oksygen, 78 % nitrogen og 1 % andre gasser.
I FreeStyle® Comfort® trekkes luften i rommet
inn i maskinen via et luftinntak. Deretter passerer
det gjennom et absorberende materiale som hetermolekylsikt. Dette materialet skiller oksygenet fra
nitrogenet og lar bare oksygenet slippe gjennom.
Resultatet er en strøm av høykonsentrert oksygen
som leveres til brukeren.
FreeStyle Comfort kombinerer avansert oksygenkonsentrator- med oksygenkonserveringsteknologier, forå skape en bærbar enhet som muliggjør maksimalbærbarhet og mobilitet. Den avanserte pulsstrømningsleveringen føler raskt når brukeren puster inn,og leverer en puls med oksygen ved begynnelsen av
hver inhalasjon. Levering med pulsstrømning er ideelt
for aktiviteter og tid hjemmefra. Den konserverer oksygenet og forlenger batteriets brukstid, i motsetning
til ved kontinuerlig strømning. FreeStyle Comfort
er lett nok til å bli båret med skulderrem. Den kan
brukes med vekselstrøm (vegg), likestrøm (bil) eller
batteristrøm.

Tiltenkt bruk
CAIRE FreeStyle Comfort-oksygenkonsentratoren
er beregnet for administrasjon av ekstra
oksygen. Enheten er ikke beregnet for å værelivsopprettholdende, og har heller ikke mulighet tilå overvåke en pasient.

Hvorfor legen din har foreskrevet
oksygenMange personer lider av forskjellige hjerte-, lunge- og
andre respiratoriske sykdommer. Et betydelig antall
av disse personene kan dra nytte av supplerende
oksygenbehandling hjemme, på reise eller når dedeltar i daglige aktiviteter hjemmefra.
Oksygen er en gass som utgjør 21 % av luften
i rommet som vi puster inn, og kroppen vår er
avhengig av en jevn forsyning for å fungere
ordentlig. Legen din foreskrev en strømningeller innstilling for å behandle din bestemte
respiratoriske tilstand.

278 - NO PN MN232-C4 L | Bruksanvisning

Selv om oksygen ikke er et vanedannende
legemiddel, kan uautorisert oksygenbehandlingvære farlig. Du må rådføre deg med lege før
du bruker denne oksygenkonsentratoren.
Utstyrsleverandøren som leverer oksygenutstyret
ditt vil demonstrere hvordan det innstilles til den
forordnede strømningshastigheten.

Operatørprofil
Konsentratorene er ment for levering av
supplerende oksygen til brukere som oppleverubehag på grunn av lidelser som påvirker
effektiviteten til lungene i overføringen av oksygeni luften til blodet. Bærbare oksygenkonsentratorer(POC-er) lagrer ikke eller inneholder ikke
oksygen. De trenger ikke å fylles opp på nytt og
kan lades opp hvor som helst hvor vekselstrøm
eller likestrøm er tilgjengelig (unntatt på et fly).
Dette gjør brukeren relativt selvhjulpen når det
gjelder hjemmebruk, bevegelighet (både innenfor
og utenfor hjemmet), mobilitet og generell livsstil.
Bruk av oksygenkonsentrator krever resept fra lege.
Den typiske brukeren av oksygenbehandlinger eldre enn 65 år og lider av en rekke
luftveissykdommer, inkludert kronisk obstruktiv
lungesykdom (KOLS). Brukerne har vanligvis
gode kognitive evner og må være i stand til å
kommunisere ubehag. Hvis brukeren ikke er
i stand til å kommunisere ubehag, eller ikke kanlese eller forstå etiketter på konsentratoren og
bruksanvisningen, anbefales det at den bare brukes
under oppsyn av noen som kan. Hvis brukere føler
noe slags ubehag under bruk av konsentratoren,skal de ta kontakt med helsepersonell. Brukernerådes til å ha ekstra oksygen tilgjengelig(dvs. sylinderoksygen) i tilfelle strømbruddeller konsentratorsvikt. Det kreves ingen andre
unike egenskaper eller brukerevner for å bruke
konsentratoren.

FreeStyle Comfort
Viktig!
Sikkerhetsinstruksjoner defineres slik:
ADVARSEL: VIKTIG SIKKERHETSINFORMASJON OM FARER SOM KAN
FORÅRSAKE ALVORLIG SKADE.
FORSIKTIG: Viktig informasjon for å
forebygge skade på FreeStyle Comfort.

ADVARSEL: ENKELTE PUSTEFORSØK FRA
PASIENTENS SIDE VIL IKKE NØDVENDIGVISUTLØSE KONSERVERINGSFUNKSJONEN.
ADVARSEL: FREESTYLE COMFORT SKAL IKKEBRUKES PÅ EN TRAKETOMISERT PASIENT.
ADVARSEL: GRAVIDE ELLER AMMENDE KVINNERSKAL IKKE BRUKE TILBEHØRENE I DENNEHÅNDBOKEN. DE KAN INNEHOLDE FTALATER.

Merk: Informasjon som trenger spesiell oppmerksomhet.

Indikasjoner for bruk
ADVARSEL: DET ER VELDIG VIKTIGÅ KUN VELGE DET FORESKREVNENIVÅET AV OKSYGEN. VALG AV OKSYGENTILFØRSEL SKAL IKKE ENDRESMED MINDRE DU HAR FÅTT BESKJEDAV AUTORISERT HELSEPERSONELLÅ GJØRE DET.
ADVARSEL: DENNE ENHETEN HAR IKKE BLITTTESTET ELLER GODKJENT FOR BRUK MEDPEDIATRISKE PASIENTER.
ADVARSEL: FØDERAL LOV (I USA) BEGRENSER
DENNE ENHETEN TIL Å SELGES TIL ELLER LEIESUT TIL EN LEGE ELLER ANNEN AUTORISERTLEVERANDØR INNEN HELSESEKTOREN.
ADVARSEL: DENNE ENHETEN SKAL IKKEBRUKES SOM LIVSOPPRETTHOLDENDE.
YTTERLIGERE OVERVÅKING KAN KREVES FORENHVER BRUKER SOM IKKE KAN KOMMUNIKEREUBEHAG MENS DU BRUKER DENNE ENHETEN.
BRUKERE MED NEDSATT HØRSEL OG/ELLER SYN
KAN HA BEHOV FOR HJELP MED Å OVERVÅKE
ALARMENE.
ADVARSEL: INNSTILLINGER FOR PULSDOSEMODUS MÅ BESTEMMES INDIVIDUELT FOR HVER
BRUKER ETTER DERES BEHOV VED HVILE, VEDFYSISK ANSTRENGELSE OG UNDER REISE.
ADVARSEL: DENNE ENHETEN ER BEREGNETFOR BRUK I HJEMMEPLEIE OG LIGNENDEPOLIKLINISKE MILJØER.

Kontraindikasjoner for bruk
ADVARSEL: I NOEN TILFELLER KANBRUK AV OKSYGEN SOM IKKE ER
FORESKREVET VÆRE FARLIG. DENNE
ENHETEN SKAL KUN BRUKES NÅR DET
ER FORESKREVET AV LEGE.
ADVARSEL: DENNE ENHETEN ER IKKEBEREGNET FOR BRUK I ELLER I NÆRHETEN
AV MAGNETRESONANSMILJØER, SLIK SOM
MR-MASKINER.
ADVARSEL: MÅ IKKE BRUKES I NÆRHETEN
AV BRANNFARLIGE ANESTESIMIDLER.
ADVARSEL: I LIKHET MED ENHVER ELEKTRISKDREVET ENHET, KAN BRUKEREN OPPLEVEPERIODER MED NEDETID SOM FØLGE
AV STRØMBRUDD ELLER BEHOV FOR
VEDLIKEHOLD AV OKSYGENKONSENTRATOREN
AV EN KVALIFISERT TEKNIKER.
OKSYGENKONSENTRATOREN PASSER IKKEFOR ENHVER BRUKER SOM KAN OPPLEVEHELSEMESSIGE BIVIRKNINGER SOM FØLGE
AV SLIKE MIDLERTIDIGE AVBRYTELSER.
ADVARSEL: DENNE ENHETEN ER IKKEGODKJENT FOR BRUK I ALLE MILJØER.
EKSPONERING FOR VISSE ELEKTROMAGNETISKEKILDER KAN RESULTERE I FORSTYRRELSER
ELLER UTILSIKTET ENHETSYTELSE. VED
FORSTYRRELSER SKAL ENHETEN STENGES
AV UMIDDELBART.

ADVARSEL: DENNE ENHETEN ER BEREGNETÅ BRUKE AV PASIENTEN SOM MOTTAROKSYGENTERAPI.

PN MN232-C4 L | Bruksanvisning NO - 279

FreeStyle Comfort
Retningslinjer for sikkerhet
ADVARSEL: INGEN FORM FOR
MODIFISERING AV DETTE UTSTYRETER TILLATT.
ADVARSEL: DENNE ENHETEN LEVEREROKSYGEN AV HØY KONSENTRASJON SOM
FREMMER RASK BRENNING. TILLAT ALDRIRØYKING ELLER BRUK AV ÅPEN ILD I SAMMEROM SOM (1) DENNE ENHETEN, ELLER (2) NOE
TILBEHØR MED OKSYGEN. DERSOM DENNE
ADVARSELEN IKKE FØLGES KAN DET FØRE TIL
ALVORLIG BRANN, MATERIELL SKADE OG/ELLERPERSONSKADE ELLER DØD.
ADVARSEL: IKKE BRUK ENHETEN I ET
AVGRENSET ELLER TRANGT ROM (F.EKS.
EN LITEN KOFFERT, ELLER EN VESKE) DER
VENTILASJONEN KAN VÆRE BEGRENSET. DETTE
KAN FØRE TIL AT OKSYGENKONSENTRATORENBLIR OVEROPPHETET OG YTELSEN SVEKKES.
ADVARSEL: KONSENTRATOREN BØR
PLASSERES SLIK AT DEN IKKE BLIR UTSATT
FOR RØYK, FORURENSNING ELLER GASSER.
ADVARSEL: HVIS DET OPPSTÅR EN ALVORLIGHENDELSE MED DENNE ENHETEN, BØR
BRUKEREN UMIDDELBART RAPPORTERE
HENDELSEN TIL LEVERANDØREN OG/ELLER
PRODUSENTEN. EN ALVORLIG HENDELSEER DEFINERT SOM EN SKADE, DØD ELLERPOTENSIAL FOR Å FORÅRSAKE SKADE/DØDDERSOM HENDELSEN SKULLE FOREKOMME
IGJEN. BRUKEREN KAN OGSÅ RAPPORTERE
HENDELSEN TIL VEDKOMMENDE MYNDIGHETI LANDET DER HENDELSEN SKJEDDE.
ADVARSEL: BRUKEN AV NOEN TYPER
TILBEHØR FOR ADMINISTRERING AV
OKSYGEN, SOM IKKE ER SPESIFISERTFOR DENNE KONSENTRATOREN, KAN
SVEKKE DENS YTELSE. ANBEFALTTILBEHØR ER VIST TIL I DENNE
BRUKERVEILEDNINGEN.
ADVARSEL: HVIS OKSYGENKONSENTRATORENHAR SUNKET, ER SKADET ELLER UTSATT FOR
VANN, KONTAKT UTSTYRSLEVERANDØREN FOR
INSPEKSJON ELLER EVENTUELL REPARASJON
AV ENHETEN. IKKE BRUK OKSYGENKONSENTRATOREN HVIS STRØMLEDNINGEN ELLERSTØPSELET ER SKADET.
ADVARSEL: NÅR OKSYGENKONSENTRATOREN
TRANSPORTERES I BIL MÅ DEN SPENNESFORSVARLIG FAST MED BELTE ELLER ANNEN
FESTEANORDNING, FOR Å UNNGÅ SKADE PÅ
ENHETEN ELLER PERSON.
280 - NO PN MN232-C4 L | Bruksanvisning

ADVARSEL: FORSKRIFTER FRA US
DEPARTMENT OF TRANSPORTATION (DOT)(TRANSPORTETATEN I USA) OG DE FORENTE
NASJONER (FN), KREVER AT BATTERIET TIL
DENNE ENHETEN FJERNES UNDER ALLE
INTERNASJONALE FLYVNINGER MED RUTEFLYNÅR OKSYGENKONSENTRATOREN SJEKKESINN SOM BAGASJE. BATTERIET MÅ OGSÅFJERNES FRA OKSYGENKONSENTRATOREN
OG PAKKES FORSKRIFTSMESSIG NÅR ENHETENSKAL SENDES.
ADVARSEL: IKKE BRUK DETTE UTSTYRETUTEN FØRST Å HA LEST OG FORSTÅTT DENNE
VEILEDNINGEN. DERSOM DU IKKE FORSTÅR
ALLE ADVARSLENE OG INSTRUKSJONENE,MÅ DU KONTAKTE UTSTYRSLEVERANDØREN
FØR DU FORSØKER Å BRUKE DETTE UTSTYRET.
HVIS DU IGNORERER DETTE, KAN DET FØRE TILPERSONSKADE ELLER MATERIELL SKADE.
ADVARSEL: OPPSØK LEGEHJELP ØYEBLIKKELIG
HVIS DU FØLER UBEHAG ELLER OPPLEVER ENMEDISINSK NØDSITUASJON.
ADVARSEL: BRUK AV OKSYGENKONSENTRATOREN VED TEMPERATURER UTENFOR DETSPESIFISERTE TEMPERATUROMRÅDET KAN
BEGRENSE KONSENTRATORENS EVNE TIL
Å AVGI DEN SPESIFISERTE OKSYGENKONSENTRASJONEN. SE ETTER TEMPERATURGRENSER
I AVSNITTET FOR SPESIFIKASJONER I DENNE
BRUKERVEILEDNINGEN.
ADVARSEL: IKKE BRUK OLJE, FETT ELLERPETROLEUM-BASERTE, ELLER ANDRE
BRENNBARE PRODUKTER SAMMEN MEDOKSYGENKONSENTRATOREN ELLER TILBEHØRMED OKSYGEN. BARE VANNBASERTE,OKSYGEN-KOMPATIBLE KREMER ELLER SALVER
SKAL BRUKES. OKSYGEN AKSELERERER
FORBRENNINGEN AV BRENNBARE STOFFER.
ADVARSEL: FEIL BRUK AV BATTERIET KAN FØRETIL AT BATTERIET BLIR VARMT, TAR FYR, OG KAN
FØRE TIL ALVORLIG PERSONSKADE. FORSIKREDEG OM AT BATTERIET IKKE STIKKES HULL
PÅ, SLÅS ELLER MISTES I BAKKEN, ELLER PÅ
ANNEN MÅTE BLIR UTSATT FOR KRAFTIGE SLAG
ELLER STØT. BRUKEN AV ET SKADET BATTERIKAN FØRE TIL PERSONSKADE.
ADVARSEL: DU MÅ IKKE KORTSLUTTE
BATTERIETS METALLKONTAKTER MED
METALLOBJEKTER, SOM NØKLER ELLERMYNTER. DETTE KAN FORÅRSAKE GNISTER
ELLER STERK VARME.

FreeStyle Comfort
ADVARSEL: NÅR OKSYGENKONSENTRATORENBRUKES UTENDØRS MED VEKSELSTRØMTILFØRSEL MÅ DEN KUN KOBLES TIL EN JORDET STIK�KONTAKT PÅ EN KURS MED JORDFEILBRYTER.

ADVARSEL: FORSØK IKKE NOE
ANNET VEDLIKEHOLD ENN DE MULIGELØSNINGENE SOM ER ANGITT I DENNE
BRUKERVEILEDNINGEN.

ADVARSEL: OKSYGENKONSENTRATOREN
BØR IKKE BRUKES RETT VED SIDEN AV ELLEROPPÅ ANNET UTSTYR. HVIS DET IKKE ER MULIG
Å UNNGÅ DETTE, BØR UTSTYRET OBSERVERESFOR Å SIKRE NORMAL DRIFT.

ADVARSEL: KUN TILBEHØR ANBEFALT AV
PRODUSENTEN. BRUK AV NOE ANNET KAN
VÆRE FARLIG, FØRE TIL ALVORLIG SKADE PÅ
OKSYGENKONSENTRATOREN OG VIL GJØREGARANTIEN UGYLDIG.

ADVARSEL: BRUK AV ANDRE KABLER OG
ADAPTERE ENN DE SOM ER SPESIFISERT,MED UNNTAK AV KABLER OG ADAPTERE SOM
SELGES AV PRODUSENTEN TIL DET MEDISINSKELEKTRISKE UTSTYRET SOM RESERVEDELER
FOR INTERNE KOMPONENTER, KAN FØRE TILØKTE UTSLIPP AV REDUSERT IMMUNITET FOROKSYGENKONSENTRATOREN.

ADVARSEL: PRODUSENTEN ANBEFALER EN
ALTERNATIV OKSYGENKILDE I TILFELLER SOM
FOR EKSEMPEL STRØMUTFALL, ALARMTILSTAND
ELLER MEKANISK FEIL. RÅDFØR DEG MED
DIN LEGE ELLER UTSTYRSLEVERANDØR
VEDRØRENDE NØDVENDIG RESERVEUTSTYR.

ADVARSEL: BRUK KUN DEN SPENNINGEN SOM
ER SPESIFISERT PÅ MERKEETIKETTEN SOM ERFESTET TIL ENHETEN.
ADVARSEL: IKKE BRUK SKJØTELEDNING
TIL DENNE ENHETEN. KOBLE HELLER IKKEFOR MANGE STØPSLER I DEN SAMMESTIKKONTAKTEN. BRUK AV SKJØTELEDNING
KAN HA EN NEGATIV EFFEKT PÅ YTELSENTIL ENHETEN. FOR MANGE STØPSLER I ÉNSTIKKONTAKT KAN FØRE TIL OVERBELASTNING
PÅ PANELET SOM KAN FØRE TIL AT SIKRINGEN
GÅR, ELLER BRANN HVIS SIKRINGEN IKKEFUNGERER SOM DEN SKAL.
FORSIKTIG: I tilfeller der du observerer en
alarm eller at oksygenkonsentratoren ikke
fungerer som den skal, se feilsøkingsdeleni denne brukerveiledningen. Rådfør deg
med utstyrsleverandøren hvis du ikkeklarer å løse problemet.
FORSIKTIG: Følg alle produsentens anvisninger for
at garantien ikke skal bli ugyldig.

ADVARSEL: PASS PÅ AT HVERKEN LUFTINNTAKET
ELLER LUFTUTTAKENE BLOKKERES. IKKESLIPP ELLER FØR NOEN GJENSTANDER INN I
ÅPNINGENE PÅ UTSTYRET. DETTE KAN FØRETIL AT OKSYGENKONSENTRATOREN BLIROVEROPPHETET OG YTELSEN SVEKKES.
ADVARSEL: RØYKING SAMTIDIG SOM BRUK AV
OKSYGEN ER DEN MEST VANLIGE ÅRSAKEN TIL
BRANNSKADER OG RELATERTE DØDSFALL. DU
MÅ FØLGE DISSE SIKKERHETSADVARSLENE:ADVARSEL: TILLAT ALDRI RØYKING, STEARINLYS
ELLER ÅPEN ILD I SAMME ROM SOM DENNE
ENHETEN, ELLER NOE TILBEHØR MED OKSYGEN.
ADVARSEL: Å RØYKE MENS MAN BRUKER ENOKSYGENKANYLE KAN FØRE TIL BRANNSKADEI ANSIKTET OG MULIG DØD.
ADVARSEL: HVIS DU RØYKER KAN DISSE
3 TRINNENE REDDE LIVET DITT: SKRU AV
OKSYGENKONSENTRATOREN, TA UT KANYLEN,
OG GÅ UT AV ROMMET DER DENNE ENHETENER PLASSERT.

PN MN232-C4 L | Bruksanvisning NO - 281

FreeStyle Comfort
ADVARSEL: SKILT MED TEKSTEN «RØYKINGFORBUDT – OKSYGEN I BRUK» MÅ VÆRETYDELIG OG SYNLIG I HJEMMET, ELLER DER
OKSYGEN ER I BRUK. PASIENTER OG DERES
PLEIERE MÅ VÆRE INFORMERT OM FARENE VEDÅ RØYKE I NÆRHETEN AV MEDISINSK OKSYGEN,
ELLER NÅR MEDISINSK OKSYGEN ER I BRUK.
ADVARSEL: IKKE BRUK OKSYGENKONSENTRATOREN I NÆRHETEN AV BRANNFARLIGE GASSER.
DETTE KAN RESULTERE I EKSPLOSIV BRANN SOM
KAN FØRE TIL MATERIELL SKADE, PERSONSKADE
ELLER DØD.
ADVARSEL: HVIS KANYLEN BLIR TATT UT OG
LAGT PÅ KLÆR, SOFAER ELLER ANDRE MYKE
MATERIALER NÅR DEN TAS UT, VIL DET KUNNE
FORÅRSAKE ØYEBLIKKELIG BRANN HVIS DETUTSETTES FOR EN SIGARETT, VARMEKILDEELLER ÅPEN ILD.
ADVARSEL: LEGG ALDRI FRA DEG ENNESEKANYLE PÅ ELLER UNDER KLÆR,SENGETØY ELLER STOLPUTER. HVIS ENHETENER SLÅTT PÅ, MEN IKKE BRUKES, VIL
OKSYGENET GJØRE MATERIALET BRENNBART.
SØRG FOR AT STRØMMEN ER AV NÅR
OKSYGENKONSENTRATOREN IKKE ER I BRUK.
ADVARSEL: GODKJENT TILBEHØR TIL
OKSYGENTILFØRSEL ER DESIGNET FOR
Å REDUSERE SPREDNINGEN AV BRANNFOR PASIENTENS SIKKERHET. BRUK AV IKKEGODKJENT TILBEHØR KAN ØKE RISIKOEN
FOR PERSONSKADE ELLER SKADE RELATERTTIL BRANN.
ADVARSEL: OKSYGENKONSENTRATORENMÅ VÆRE AV OG KOBLE FRA EKSTERN STRØM
FØR DU FJERNER BATTERIPAKKEN.
ADVARSEL: SMÅ DELER KAN
UTGJØRE EN KVELNINGSFARE.
ADVARSEL: UNNGÅ KONTAKT MED UTLUFTINGENI MER ENN 10 SEKUNDER, DA DET KAN VÆREVARMT.
ADVARSEL: VIND ELLER STERK TREKKKAN PÅVIRKE NØYAKTIG LEVERING AV
OKSYGENBEHANDLING.
ADVARSEL: DETTE PRODUKTET KAN EKSPONEREDEG FOR KJEMIKALIER, INKLUDERT NIKKEL,
SOM IFØLGE STATEN CALIFORNIA KANFORÅRSAKE KREFT. FOR MER INFORMASJON,GÅ TIL WWW.P65WARNINGS.CA.GOV.

282 - NO PN MN232-C4 L | Bruksanvisning

FORSIKTIG: Koble alltid fra
strømtilførselen fra stikkontakten
i veggen før kontakten på
oksygenkonsentratoren kobles fra.
FORSIKTIG: Ikke plasser oksygenkonsentratoren slik
at det er vanskelig å få tilgang til strømledningen.
FORSIKTIG: Plasser alltid oksygenforsyningsslangenog strømledningene på en måte som forebygger
snublefare eller mulig utilsiktet kvelning.
FORSIKTIG: Når oksygenkonsentratoren brukes
i bil, båt eller et annet sted med likestrømtilførsel,
sørg for at kjøretøyet startes opp før oksygenkonsentratoren kobles til. Dersom disse instruksjonene
ikke følges, kan det resultere i at strømtilførselen
ikke gir strøm til oksygenkonsentratoren.
FORSIKTIG: Når bilen som du bruker oksygenkonsentratoren i er slått av, kobler du fra og fjernerenheten fra bilen. Oppbevar aldri oksygenkonsentratoren i en veldig varm eller kald bil, eller i andre
lignende miljøer med høy eller lav temperatur.
Merk: Bærbart og mobilt RF-kommunikasjonsutstyr
kan påvirke medisinsk elektrisk utstyr.
FORSIKTIG: La IKKE oksygenkonsentratoren eller strømtilførselen være tilkoblet
kjøretøyet hvis tenningen er slått AV. Det
kan tømme bilens batteri for strøm.
FORSIKTIG: Drift eller oppbevaring av
oksygenkonsentratoren utenfor dens normale
temperaturområde og omgivelsestrykkområde
(høyde) kan påvirke ytelsen og senke driftstiden
for batteriet, og/eller øke ladetiden. Se avsnittet
for Spesifikasjoner i denne bruksanvisningen fortemperaturgrenser for lagring og drift.
FORSIKTIG: For oksygenkonsentratorer som
er utstyrt med batterier: Lagres kjølig og tørt
for å sikre lang levetid for batteriet. Lagring av
oksygenkonsentratoren over lengre tid ved høy
temperatur eller med et fulladet eller fullstendig tomt
batteri kan redusere batteriets samlede levetid. Ikke
prøv å åpne batteriet. Det er ingen innvendige delersom kan repareres. Hold batteriene unna barn.
FORSIKTIG: BRUK KUN batterier levert av
produsenten. For riktig avhending av batteriene,kontakt utstyrsleverandøren eller lokale
myndigheter for avhendingskrav.

FreeStyle Comfort
Merk: For å redusere skader forårsaket av konsentratoren
eller strømbrudd, bør utstyrsleverandøren vurdere
behovet for reserveoksygen under installasjonen og med
jevne mellomrom når disse egenskapene endres:- Tilstanden til pasienten- Miljøet pasienten lever i- Evnen til å forsyne pasienten med
reserveforsyninger av supplerende oksygen.
Merk: Påse at kanylen sitter helt inne og godt fast.
Dette sikrer at oksygenkonsentratoren kan utføre
korrekt registrering av innånding for oksygentilførsel.
Ved innånding bør du høre eller føle oksygen flyte
til utstikkerne på nesekanylen.

Merk: Hvis oksygenkonsentratoren har vært lagret
i en lengre periode utenfor området for normal
driftstemperatur som er vist i tabellen ovenfor,skal du la enheten komme tilbake til normal
driftstemperatur før den slås på.
Merk: Utstyrsleverandøren må vurdere behovet
for reserveoksygen i tilfelle konsentratorsvikt eller
strømbrudd ved installasjon og med jevne mellomrom
når disse attributtene endres:- Tilstanden til pasienten- Miljøet pasienten lever i- Evnen til å forsyne pasienten med
reserveforsyninger av supplerende oksygen.

Merk: Litiumbatterier kan permanent miste sinkapasitet når de eksponeres for ekstremt varme
temperaturer med batteriene fullt oppladet eller
helt utladet. Ved langvarig lagring anbefales det atbatteriene er ladet 25 til 50 % og oppbevares i ettemperaturområde på -20 °C – 20 °C (-4 °F – 68 °F).
Merk: Ikke bruk oksygenkonsentratoren uten at
luftinntaksfilteret er på plass. Hvis enheten er levert
med et ekstra filter, sett inn dette «reservefilteret» før
du rengjør det skitne filteret. Rengjør det skitne filtereti varmt vann med såpe, tørk deretter grundig før bruk.
Merk: Produsenten anbefaler ikke sterilisering
av dette utstyret.
Merk: Det er aldri fare for å utarme oksygenet
i et rom hvor du bruker oksygenkonsentratorenheten.
ADVARSEL: OKSYGENKONSENTRATOREN SKAL IKKE BRUKES ELLER
KOBLES TIL EKSTERNSTRØM (AC/DC)UTEN AT ET BATTERI ER SATT INN.
IKKE FORSØK Å SLÅ PÅ ENHETEN ELLER KOBLE DEN TIL EKSTERN STRØM
NÅR BATTERIPAKKEN ER FJERNET.
BATTERIPAKKEN SKAL BARE FJERNES NÅRENHETEN ER SLÅTT AV OG ER KOBLET FRASTRØMFORSYNINGEN. IKKE FORSØK Å FJERNEBATTERIPAKKEN MENS ENHETEN FORTSATTER PÅ ELLER KOBLET TIL STRØMFORSYNINGEN.

PN MN232-C4 L | Bruksanvisning NO - 283

FreeStyle Comfort
Spesifikasjoner
FreeStyle Comfort
Modeller (med ett batteri)10.0 x 7.3 x 3.1 in (25.4 x 18.5 x 7.9 cm)

Mål (med dobbelt batteri)11.0 x 7.3 x 3.1 in (27.9 x 18.5 x 7.9 cm)

Vekt (med én batteripakke)

5 lbs (2,3 kg)

Vekt med (dobbel batteripakke)

6 lbs (2,7 kg)

Nominelt lydnivå***Innstillinger for strømningO2-effektinnstilling([ved 20 bpm STPD] Alle volumer +/- 15 %)Oksygenkonsentrasjon*

39,93 dB(A) ± 0,42 dB(A) ved 2
Pulsinnstillinger 1, 2, 3, 4, 5
Innstilling 1–210 ml/min
Innstilling 2–420 ml/min
Innstilling 3–630 ml/min
Innstilling 4–840 ml/min
Innstilling 5–1050 mL/min
90 % (+5.5 % / -3 %)Oksygenovervåkning

Ja

Maksimalt utløpstrykk<30 PSI

Doseringsfølsomhet> -0.1 cm H2O (justerbar)

Vekselstrøm
Likestrøm

100–240 VAC, 50–60 Hz, 120 W maks
11–18 VDC (10 maks ampere)

Batteritype Litiumion
Batterikapasitet
Batteriladetid
Batteripakkens varighet
Driftstemperatur**

Enkeltcelle: 6700 mAh; Dobbeltcelle: 13400 mAh
Enkeltbatteri: 3,5 timer
Dobbelt batteri: 6,0 timer
Varighet av enkeltbatteri: Innstilling 2-inntil 4 timer
Varighet av dobbelt batteri: Innstilling 2-inntil 8 timer
5 °C til 40 °C (41 °F til 104 °F)

Transports- og oppbevaringstemp. (Enhet)-25 °C til +70 °C (-13 °F til +158 °F) opp til 3048 m (10000 fot)Transports- og oppbevaringstemp. (Batteri)-25 °C til +70 °C (-13 °F til +158 °F) opp til 3048 m (10000 fot)

Høyde ved drift**

3048 m til -381 (-1250 til 10000 fot) (testet til 700 – 1060 hPa)

Luftfuktighet ved drift
Luftfuktighet ved oppbevaring
Begrenset garanti

15–95 % relativ luftfuktighet (ikke-kondenserende)
0–90 % ikke-kondenserende
Se garantierklæring

* Basert på et atmosfærisk trykk på 14,7 psi (101 kPa) ved 21 °C (70 °F)** Enheten vil fungere i alle miljøer så lenge miljøet er med spesifiserte temperatur- og fuktighetsområder. Drift utenfor
disse driftsspesifikasjonene kan begrense konsentratorens evne til å oppfylle oksygenkonsentrasjonsspesifikasjonen
ved høyere liters strømningshastigheter. Oppvarmingsskjermen kan bli vist vedvarende til konsentratoren har nåddmålkonsentrasjonen eller opptil 15 minutter. Fortsett å bruke enheten i løpet av denne tiden.
*** Lydnivå målt per testmetode Nr. 14-1 10/2018 MDS-Hei.
Forventet levetid er minst fem år.

284 - NO PN MN232-C4 L | Bruksanvisning

FreeStyle Comfort
Instruksjoner for bruk
1. Finn og plasser FreeStyle Comfort på etpassende sted for bruk.
a. Konsentratoren skal være godt ventilert
slik at luftinntaks- og luftuttaksfiltrene ikke
er blokkert.
b. Konsentratoren skal posisjoneres slik
at alle hørbare og visuelle indikatorer
og alarmer er lette å se og høre.
c. Sørg for at luftinntaksfiltrene er på plass
før du bruker enheten. Ta kontakt med
utstyrsleverandøren din hvis enheten
mangler et filter.

3. Plugg inn strømledningen for veksel- eller
likestrøm i strømkontakten til høyre på forsiden
av enheten.

2. Installer en batteripakke i konsentratoren.
a. Juster batteriet med bunnhuset fra midten
av FreeStyle Comfort.

4. Koble en nesekanyle til oksygenuttakets hake.
Sørg for at koblingen sitter godt fast før drift.
MERK: Bruk av andre kanyler enn de som er anbefalt
kan påvirke ytelsen.

Merk: Til utstyrsleverandøren: Følgende tilbehør til
administrering av oksygen anbefales å bruke med
FreeStyle Comfort:b. Skyv batteriet innover til du hører et klikk
for å bekrefte at det er låst på plass.

• Nesekanyle med en slange på ca. 2,1 m (7 fot)(maksimum LPM: 6 LPM): CAIRE delenummer
5408-SEQ• Ytterligere informasjon om anbefalt tilbehør
er tilgjengelig online på www.caireinc.com.
5. Slå på FreeStyle Comfort ved å trykke
på på/av-knappen og velg foreskrevet
strømningsinnstilling ved å bruke knappene
for økning (+) eller reduksjon (-) av
oksygentilførsel.

PN MN232-C4 L | Bruksanvisning NO - 285

FreeStyle Comfort
Pustefrekvens
15
20
25
30
35
40

Innstilling for ikke-autoSat-pulsvolum (ml)*
1
2
3
4
5
14.0
10.5
8.4
7.0
6.0
5.3

28.0
21.0
16.8
14.0
12.0
10.5

42.0
31.5
25.2
21.0
18.0
15.8

56.0
42.0
33.6
28.0
24.0
21.0

70.0
52.5
42.0
35.0
30.0
26.3

En alarm vil lyde kort for å indikere at enheten
er slått på.

8. Slå av FreeStyle Comfort ved å trykke påpå/av-knappen.

Merk: Når FreeStyle Comfort slås på for først gang,er standard oksygeninnstilling 2. Hvis innstillingen
endres, aktiveres enheten i oksygeninnstillingen som
ble brukt sist.

Merk: Skjermen vil gå inn i standbymodus og dimmes
etter 20 sekunder. Trykk på enhver knapp for å vekke
skjermen fra standbymodus.

6. Posisjoner nesekanylen over ansiktet i henhold
til produsentens instruksjoner levert med
kanylen, og pust normalt. Du skal føle et pust av
oksygen levert gjennom nesekanylen hver gang
du puster inn.
7. For å endre oksygeninnstillingen, trykk på
knappene for økning (+) eller reduksjon av (-)
oksygentilførsel for å velge ønsket nivå. Det er
normalt å høre en forskjell i lyd når du endrer
strømningsinnstillinger.

9. For avanserte funksjoner, se instruksjoner ommenynavigering.

SPESIELLE PRODUKTEGENSKAPER
autoSAT®
CAIREs proprietære autoSAT-funksjon lar
FreeStyle Comfort levere et konsistent volum
av oksygen (bolus) per pust, opp til en maksimalpustefrekvens. Ettersom brukerens pustefrekvensøker under aktiviteter som å gå, justerer autoSATfunksjonen enhetens drift for å opprettholde
et innstilt bolusvolum.
UltraSense®
Denne sensitive pulsdoseutløseren oppdager
trykkendringen når du puster, og frigjør oksygennår du inhalerer.

Pustefrekvens
15
20
25
30
35
40

autoSAT-pulsvolum (mL)*Innstilling
1
2
3
4
5
10.5
10.5
10.5
10.5
10.5
10.5

21.0
21.0
21.0
21.0
21.0
21.0

31.5
31.5
31.5
31.5
30.0
26.3

42.0
42.0
42.0
35.0
30.0
26.3

*Toleranse er +/- 15 %

286 - NO PN MN232-C4 L | Bruksanvisning

52.5
52.5
42.0
35.0
30.0
26.3

autoDOSE®
Hvis en «ingen pust-alarm» oppstår, vil denne
funksjonen pulsere oksygen ved 20 pust per minutt
ved enhetens gjeldende strømningsinnstilling til
alarmtilstanden er løst.

FreeStyle Comfort
Strømforsyninger
FreeStyle Comfort kan motta strøm på tre måter:• Oppladbar batteripakke• Vekselstrøm• Likestrøm
Informasjon om driften av hver av disse
komponentene er i følgende avsnitt.
ADVARSEL: BRUK KUN STRØMFORSYNINGENE SOM ER BESKREVETI DENNE BRUKSANVISNINGEN. BRUKAV STRØMFORSYNINGER OG BATTERIPAKKER SOM IKKE ER FRA CAIREKAN FØRE TIL SKADE PÅ ENHETEN
OG FEILAKTIG DRIFT.

FJERNE BATTERIET

Trykk batteriutløserknappen nedover, vekk fra
FreeStyle Comfort. Mens du HOLDER knappen,
skyver du batteriet ut ved å dra det vekk fra
konsentratoren.
ADVARSEL: OKSYGENKONSENTRATOREN MÅ VÆRE AV OG KOBLETFRA EKSTERN STRØM FØR FJERNING
AV BATTERIPAKKEN.

BATTERIPAKKE

FreeStyle Comfort kan drives av den oppladbare
batteripakken som følger med enheten. Det er ingen
driftsbegrensninger for konsentratoren når den brukes
med batteripakken. Alle strømningshastigheter ertilgjengelige på batteristrøm.

KONTROLLERE LADINGEN
AV BATTERIPAKKENINSTALLERE BATTERIET

Juster batteriet med bunnhuset fra midten av
FreeStyle Comfort. Skyv batteriet innover til duhører et klikk for å bekrefte at det er låst på plass.

For å kontrollere batteriladningen når batteriet
er installert i FreeStyle Comfort, se batteriikonet
og den gjenværende tiden på displayet.
80–100 %60–80 %

40–60 %

20–40 %20 %PN MN232-C4 L | Bruksanvisning NO - 287

FreeStyle Comfort
Merk: Hvis strømforsyningen er koblet til, vil en pluggindikator vises i stedet for ladeprosenten som gjenstår.
For å kontrollere batteriladingen når den ikke er
installert i FreeStyle Comfort, legg den flat og trykk
på TEST-knappen. Indikatorlyset for batterimåler
(25–100 %) vil tennes til høyre for TEST-knappenfor å indikere nivået på lading av batteripakken.

VARIGHET BATTERIDRIFT

Når ekstern vekselstrøm eller likestrøm kobles fra,vil konsentratoren automatisk skifte til å bruke
batteristrøm hvis et ladet batteri er installert.
Tabellen nedenfor viser typisk varighet for en ny
batteripakke. Varigheten til batteripakken holder
seg konstant, uavhengig av pustehastigheten din.
Innstilling av
puls

Varighet av
enkeltbatteri(8 celler)

Varighet av dobbelt batteri(16 celler)

1

opp til 8 timer

opp til 16 timer

2

opp til 4 timer

opp til 8 timer

3

opp til 3 timer

opp til 6 timer

4

opp til 2,25 timer

opp til 4,5 timer

5

opp til 2 timer

opp til 4 timer

TYPISK BATTERILADNINGSTID
FØRSTE LADING

Det nye batteriet som følger med din FreeStyle
Comfort er bare delvis ladet når det sendes
fra fabrikken. Før første gangs bruk skal dulade batteripakken helt opp med bruk av enten
vekselstrøm eller likestrøm.

GENERELL INFORMASJON LADINGFreeStyle Comfort vil lade opp batteripakken
hver gang den installeres i konsentratoren og
konsentratoren er koblet til en ekstern vekselstrømeller likestrømkilde (unntatt på et fly).
Batteriet er fulladet når det viser 100 %.
Alternativt kan batteripakken lades ved brukav den valgfrie bordladeren.

288 - NO PN MN232-C4 L | Bruksanvisning

Fullstendig oppladning av en batteripakke med
et fullstendig utladet batteri er 3,5 timer for et
enkeltbatteri og 6,0 timer for et dobbeltbatteri.
Merk: Batteriet vil kanskje ikke bli ladet med en ganghvis batteriet er fullstendig utladet.

BATTERILEVETIDFreeStyle Comfort-batterier er utformet for å varei minst 300 oppladnings-/utladningssykluser.

FreeStyle Comfort
Veksel- og likestrømtilførsel
FreeStyle Comfort leveres med vekselstrømsog likestrømstilførsel som standard.
Vekselstrømtilførselen inkluderer følgende:	a. Vekselstrømkloss som opererer mellom
100‑240 volt, 50-60 Hz.
	b. Vekselstrømledning for å koble
strømklossen til tilsvarende strømuttak.

Likestrømtilførselen består av en ledning som
plugges direkte inn i enheten og likestrømuttaket.

VEKSELSTRØM

2. Koble vekselstrømledningen forvekselstrømklossen.

Vekselstrøm er til bruk hjemme eller alle stederhvor standard stikkontakter med vekselstrøm er
tilgjengelig. FreeStyle Comfort er fullt ut funksjonell
med vekselstrøm. FreeStyle Comfort vil også lade opp
batteripakken hver gang vekselstrøm er tilgjengelig(unntatt på et fly). Batteripakken vil lades opp igjen
uansett om FreeStyle Comfort er slått på eller av.
ADVARSEL: IKKE KOBLE TIL
EKSTERN STRØM UTEN AT EN
BATTERIPAKKE ER INSTALLERTI OKSYGENKONSENTRATOREN.
For å koble FreeStyle Comfort til vekselstrøm:
1. Koble vekselstrømklossen til strømkontakten
nederst på forsiden av FreeStyle Comfort.
Sørg for å trykke innover, slik at koblingen
er godt festet.

3. Koble den motsatte enden av strømledningenfor vekselstrøm til stikkontakten i veggen eller
en annen passende stikkontakt.

PN MN232-C4 L | Bruksanvisning NO - 289

FreeStyle Comfort
4. Slå på FreeStyle Comfort ved å trykke på på/avknappen og velg foreskrevet strømningsinnstillingved å bruke knappene for økning (+) eller
reduksjon (-) av oksygentilførsel.

4. Slå på FreeStyle Comfort ved å trykke på på/avknappen og velg foreskrevet strømningsinnstillingved å bruke knappene for økning (+) eller
reduksjon (-) av oksygentilførsel.
Merk: Hvis batteripakken må lades opp, vil batteriikonet på displayet vise en eneste oransje stolpe.

LIKESTRØM

Likestrøm brukes i utstyrskontakter i biler, båter
eller andre motorkjøretøy. FreeStyle Comfort
er fullt ut funksjonell med likestrøm. FreeStyle
Comfort vil også lade opp igjen batteripakken,
hvis nødvendig, hver gang likestrøm er tilgjengelig.
Batteripakken lades opp, både når enheten er av
eller på og i bruk.
ADVARSEL: IKKE KOBLE TIL
EKSTERN STRØM UTEN EN
BATTERIPAKKE INSTALLERTI OKSYGENKONSENTRATOREN.
1. Start motorkjøretøyet.
2. Koble likestrømforsyningen til strømkontakten
nederst til høyre på forsiden av FreeStyle
Comfort. Sørg for å trykke innover, slik
at koblingen er godt festet.

3. Koble den motsatte enden av
likestrømsledningen til kontakten i bileneller en annen utstyrskontakt for likestrøm.
290 - NO PN MN232-C4 L | Bruksanvisning

FORSIKTIG: For å forhindre mulig branneller overoppheting, sørg for at 12 V
DC-strømuttaket er fritt for rusk og at
adapterpluggen til DC-strømforsyningen
passer ordentlig og forblir sikkert på plass.
FORSIKTIG: DC-strømforsyningens metalloverflaterkan bli VELDIG VARME når de er i bruk på grunn
av høy strøm. La overflatene avkjøles før duberører dem.
FORSIKTIG: Ikke bruk likestrømforsyningen med
noen skjøteledninger, splittere eller adaptere.
Bruk av ikke-godkjent tilbehør kan føre til at det
overopphetes eller blir skadet.
FORSIKTIG: Å koble enheten til en 12 V DCstrømuttaket kan tømme kjøretøyets batteri hvis
motoren ikke går, selv om enheten er slått av.
FORSIKTIG: Ikke start et kjøretøy mens
DC-strømforsyningen er tilkoblet.

FreeStyle Comfort
Rengjøring, pleie og rutinemessig
vedlikehold
RUTINEMESSIG VEDLIKEHOLDForebyggende vedlikehold
Enheten er vedlikeholdsfri. Filtre inne i enheten er
ment å vare hele levetiden til enheten. Hvis enheten
ikke fungerer som den skal, bør enheten returneres
til leverandøren for reparasjon. Vedlikehold av
indre komponenter inne i kabinettet på FreeStyle
Comfort må utføres av oksygenleverandøren.
Operatøren er ansvarlig for følgende
vedlikeholdsprosedyrer.
• Ukentlig – Rengjøre luftinntaksfiltere
ADVARSEL: IKKE FORSØK Å ÅPNEELLER FJERNE PLASTDEKSLENETIL OKSYGENKONSENTRATOREN.
ENHETEN SKAL KUN ÅPNES AV
EN KVALIFISERT SERVICETEKNIKER.

LUFTINNTAKSFILTERE

Luft går inn i FreeStyle Comfort gjennom etluftinntaksfilter plassert nederst på forsiden av
oksygenkonsentratoren. Dette filteret hindrer støv
og andre store partikler i luften fra å komme inn
i enheten.
Luftinntaksfiltrene skal rengjøres minst éngang i uken. Oksygenleverandøren vil kanskjeråde deg til å rengjøre dem oftere, avhengig av
driftsforholdene. Følg de følgende trinnene for
riktig rengjøring av luftinntaksfiltrene.
1. Fjern de to filtrene fra forsiden av FreeStyle
Comfort ved å trykke opp på den tilsvarende
tappen og trekke den utover.

ADVARSEL: IKKE BRUK OKSYGENKONSENTRATOREN UTEN AT LUFTINNTAKSFILTERET ER PÅ PLASS.

VEDLIKEHOLD AV BATTERIPAKKE

Ikke la batteripakken bli eksponert for væsker.
Ikke forsøk å demontere eller åpne batteripakken.
Ikke eksponer batteriet for åpne flammer.
Ikke mist batteripakken.
Oppbevar batteripakken utenfor barns rekkevidde.
Hvis du ser at batteripakken virker skadet,ta kontakt med oksygenleverandøren før bruk.
Ikke plasser batteriet i eller i nærheten av en
mikrobølgeovn eller andre apparater for matlaging.

LAGRING AV BATTERIPAKKEN

Batteriet til oksygenkonsentratoren trenger
ikke å lades helt ut før den lades opp
igjen. Det anbefales å lade opp batteriet tiloksygenkonsentratoren etter hver bruk.
Ikke legg batteriet på et sted som eksponeresfor sterk varme, som i direkte sollys eller i enbil. Dersom dette gjøres kan det svekke batteriets
ytelse og brukstid. Det kan også føre til at batteriet
genererer varme, røyk, tar fyr eller eksploderer.
Batteripakken skal lagres under stabile forhold,kjennetegnet av lav fuktighet (mindre enn 70 %RF) og fritt for etsende gasser. For å forhindrerust skal du unngå forhold som kan skape
kondensering, som for eksempel raske svingninger
i omgivelsestemperaturen.
Dersom batteriet skal lagres i en måned eller
mer, anbefales det at batteriet er helt utladet eller
delvis oppladet mellom 50 % og 75 % (to av tre
LED-er). Når det er oppladet fjerner du batteriet
fra oksygenkonsentratoren.

2. Vask begge filtrene i en varm løsning av såpeog vann.
3. Skyll filtrene nøye og fjern overflødig vann med
en myk, lofri klut. Forsikre deg om at filtrene
er helt tørre før du setter det på plass igjen.
4. Installer de rene, tørre filtrene på nytt.
PN MN232-C4 L | Bruksanvisning NO - 291

FreeStyle Comfort
RENGJØRING

Slå AV enheten og koble fra vekselstrøm eller
likestrøm før enhver rengjøring. IKKE spraydirekte på det ytre dekselet. Bruk en fuktig (ikkevåt) klut eller svamp. Spray kluten eller svampen
med en mild husholdningsrengjøringsløsning forå rengjøre kabinettet og strømforsyningene. Fortsett
som angitt av produsenten av rengjøringsmiddelet.
ADVARSEL: FOR Å FOREBYGGE UTILSIKTET ELEKTRISK STØT OG BRANNFARE, SLÅ AV ENHETEN OG KOBLESTRØMLEDNINGEN FRA STIKKONTAKTEN FØR DU RENGJØR ENHETEN. DET
ER BARE UTSTYRSLEVERANDØRENELLER EN KVALIFISERT TEKNIKER
SOM SKAL FJERNE DEKSLENE ELLER
REPARERE ENHETEN.
ADVARSEL: UTVIS FORSIKTIGHET FOR Å
FORHINDRE AT OKSYGENKONSENTRATOREN
OG BATTERIPAKKEN BLIR VÅT. VÆSKER MÅ IKKE
KOMME INN I ENHETEN. DETTE KAN FØRE TIL EN
FUNKSJONSFEIL ELLER AT ENHETEN SLÅR SEG
AV, OG FØRER TIL ØKT RISIKO FOR ELEKTRISKSTØT ELLER BRANNSKADER.
ADVARSEL: IKKE BRUK VÆSKE DIREKTEPÅ ENHETEN. EN LISTE OVER UØNSKEDEKJEMISKE MIDLER INKLUDERER, MEN ERIKKE BEGRENSET TIL, FØLGENDE: ALKOHOLOG ALKOHOLBASERTE PRODUKTER,
KONSENTRERTE KLORBASERTE PRODUKTER(ETYLENKLORID), OG OLJEBASERTEPRODUKTER. DISSE SKAL IKKE BRUKES
TIL Å RENGJØRE PLASTHUSET PÅ
KONSENTRATOREN, DA DE KAN SKADE
ENHETEN.
ADVARSEL: KABINETTET, KONTROLLPANELETOG STRØMLEDNINGEN SKAL BARE RENGJØRES
MED ET MILDT VASKEMIDDEL PÅFØRT MED ENFUKTIG (IKKE VÅT) KLUT ELLER SVAMP, OG
DERETTER SKAL ALLE OVERFLATES TØRKES
MED EN TØRR KLUT. IKKE LA NOE SLAGS VÆSKE
KOMME INN I KONSENTRATOREN. VÆR SPESIELTNØYE MED OKSYGENUTLØPET FOR KANYLETILKOBLINGEN SLIK AT DETTE HOLDES FRI FOR
STØV, VANN OG PARTIKLER.

292 - NO PN MN232-C4 L | Bruksanvisning

ADVARSEL: BRUK EN FUKTIG (IKKE VÅT)KLUT ELLER SVAMP TIL Å RENGJØREBATTERIPAKKEN. SPRAY FØRST ET MILDTVASKEMIDDEL PÅ KLUTEN ELLER SVAMPEN,
OG RENGJØR DERETTER UTSIDEN AV
BATTERIPAKKEN.
ADVARSEL: IKKE RENGJØR MED AUTOMATISKVASKESYSTEM.

UTSKIFTING AV KANYLE

Følg alltid kanyleprodusentens anvisninger for riktig
bruk. Bytt nesekanylen eller oksygenslangen hver
3. til 6. måned (hver 3. måned ved daglig bruk)i henhold til produsentens anbefaling som anbefalt
av kanyleprodusenten eller oksygenleverandøren.
Din lege eller oksygenleverandør vil gi deg
informasjon om rengjøring og utskifting.
Ytterligere materiell for utskifting er tilgjengelig
gjennom din oksygenleverandør.

FreeStyle Comfort
Alarmtilstander
Prioritetsnr.

Alarmbeskrivelse

1

Systemfeil,
høy batteritemperatur

Display

Lydsignal

Flere detaljer / Foreslått tiltak

1 pipetone hvert
20. sek.

Enheten vil avgi en alarm når batteritemperaturen nærmer seg terskelen for
sikkerhetsavbrudd.
Enheten slår seg av når terskelen er
nådd. Hvis ekstern strøm er tilkoblet, vilalarmskjermen vises og lydalarmen fortsette i 2 minutter.
Test med batteri ved romtemperatur. Kontakt leverandør hvis problemet vedvarer.

2

Systemfeil,
høy
temperatur

1 pipetone hvert
20. sek.

Enheten vil avgi en alarm når den interne
temperaturen nærmer seg terskelen for
sikkerhetsavbrudd.
Hvis terskelen er nådd, vil enheten slå seg
av, alarmskjermen vises og lydalarmen
fortsette i 2 minutter.
Kontroller at det ikke er noen blokkeringer
på enhetens luftehull. Kontakt leverandør
hvis problemet vedvarer.

3

Systemfeil
høyt trykk(høyt systemtrykk eller
kompressorfeil)

1 pipetone hvert
20. sek.

4

Systemfeil,
oksygentilførsel (sviktende
OCD-ventil,
blokkering av
gassbane)

1 pipetone hvert
30. sek.

Kontakt leverandør hvis problemetvedvarer.

5

Driftstans
ved lavt
batterinivå

1 pipetone hvert
30. sek.

Alarmen lyder til batteriet går tomt for
strøm og enheten slår seg av.

Hvis terskelen er nådd, vil enheten slå seg
av, alarmskjermen vises og lydalarmen
fortsette i 2 minutter.
Kontakt leverandør hvis problemetvedvarer.

Lad batteriet.

PN MN232-C4 L | Bruksanvisning NO - 293

FreeStyle Comfort
Prioritetsnr.

Alarmbeskrivelse

6

Ingen pust
oppdaget

Display

Lydsignal

Flere detaljer

1 pipetone hvert
10. min

Ingen pust oppdaget på 15 sekunder.
Aktivert etter første pust eller når
oppvarmingsskjerm slås av, avhengig av
hva som skjer før.
Sjekk kanylen for knekker eller lekkasjer.
Øk innstillingen for følsomhet. Kontakt
leverandør hvis problemet vedvarer.

7

Oppvarming

ingen

Enheten starter med automatisk dosering
til et pust oppdages eller oppvarming
slutter, avhengig av hva som skjer først.

8

Lav
konsentrasjon

1 pipetone hvert
10. min,i opp til
30. minKontakt leverandør hvis problemetvedvarer.

9

Varsel ved
lavt batterinivå

1 pipetone,
gjentasikke

Lad batteriet.

10

Pustefrekvens
overskredet

1 pipetone,
gjentasikke

Pustefrekvens overskrider 40 bpm.

11

Systemfeil,
viftefeil

ingen

Kontakt leverandør hvis problemetvedvarer.

12

Systemfeil,lavt trykk

ingen

Kontakt leverandør hvis problemetvedvarer.

13

Systemfeil,
lav temp.

ingen

La enhet varmes opp til romtemperatur. Kontakt leverandør hvis problemetvedvarer.

294 - NO PN MN232-C4 L | Bruksanvisning

Hvis dosen utløses ved en feil, må duredusere følsomhetsinnstillingen. Kontakt
leverandør hvis problemet vedvarer.

FreeStyle Comfort
Prioritetsnr.

Alarmbeskrivelse

Info

Oppstart

Display

Lydsignal

Flere detaljer

1 infopip

Skjermen vises når enheten slås på, og vil
ett avgi pip for å bekrefte at summeren
fungerer.
Oppstartsskjermen viser gjeldende
fastvarerevidering nederst til høyre.

Info

Normalkjøreskjerm i
autoSAT-modus (ingenlikestrøm
tilkoblet)

ingen

Info

Normalkjøreskjerm
med autoSAT
deaktivert(ingenlikestrøm
tilkoblet)

ingen

Info

Innstilling
endret1 infopip

Info

Plugg innekstern strøm
i avslått
tilstand

ingen

Info

Plugg innekstern strøm
i driftstilstand

ingen

Info

Koble fraekstern strøm
i avslått
tilstand1 infopip

Viser CAIRE-skjermen i 2 sekunder,
etterfulgt av batteri- og støpselstatusi 5 sekunder før skjermen slås av.

Vis CAIRE-skjermen i 2 sekunder og slå
deretter skjermen av.

PN MN232-C4 L | Bruksanvisning NO - 295

FreeStyle Comfort
Prioritetsnr.

Alarmbeskrivelse

Display

Info

Koble fraekstern strøm
i driftstilstand(med batteri
til stede)

1 infopip

Info

I standbymodus – når
opp- eller
ned-knappen
trykkes, viser
displayet
batteri- eller
ladetilstanden

ingen

Info

Flymodus på
batteristrøm

1 langvarig pip

Info

Flymodus med
strømledning
tilkoblet

1 langvarig pip

296 - NO PN MN232-C4 L | Bruksanvisning

Lydsignal

Flere detaljer

Skjermen vises 5 sek. før den slukkes.

FreeStyle Comfort
Instruksjoner for menynavigering
Du kan gå inn i menyen i Ventemodus eller
kjøremodus. I kjøremodus lukkes menyen
automatisk hvis en alarmtilstand oppstår.
Hvis ingen knapper trykkes i 20 sekunder, avsluttes
menyen automatisk. I ventemodus må du kanskje
trykke på menyknappen to ganger for å vekke
enheten, og deretter gå inn i menyen.
Der er fire menyskjermer:
1. Statusskjerm: Viser revisjonsnummer, kjøretid,
batteriutladningssykluser for fastvare

Flymodus

2. Flymodus (PÅ/AV)
3. Følsomhet, justerbar (1-5)
4. autoSAT (PÅ/AV)

Følsomhet

Eksempel på statusskjerm

BRUK AV INNSTILLINGSMENYDu går inn i innstillingsmenyen ved å trykke
på Meny/Select-knappen
standbymodus.

autoSATi kjøre- eller

For å rulle gjennom menyene, trykk på.

StatussidePN MN232-C4 L | Bruksanvisning NO - 297

FreeStyle Comfort
STATUSSIDE
Statussiden til FreeStyle Comfort viser
fastvarerevisjon, driftstimer og antall
batterisykluser på det tilkoblede batteriet.

For å avbryte Flymodus på, trykk på
AVBRYT-ikonet er synlig.

mens

Flymodus på bekreftet

Statusside

Kjøretimer vises i hele timer.
Batteriutladninger er antall sykluser det tilkoblede
batteriet har blitt tømt og ladet opp igjen.

FLYMODUS
FreeStyle Comfort leveres med flymodus slått av.

For å slå av flymodus, gjenta skrittene ovenfor
mens du veksler til AV.
Trykk pånår skjermen viser «Neste»-symboletfor å flytte menysiden til neste skjermalternativ.
Flymodus er egnet for bruk i kjøremodus.
Flymodus deaktiveres automatisk hvis enheten blirværende i Ventemodus på-batteristrøm.
Merk: Se avsnittet Aktiver flymodus hvis du vil vite
mer om flymodus.

FØLSOMHETSMODUS

Hovedskjerm i flymodus

For å slå modusen på, trykk på
mens
ENDRE-ikonet er synlig. Verdien vil skifte til det
blå ordet «PÅ».

FreeStyle Comfort leveres med en standard
følsomhetsmodusinnstilling på 2 fra fabrikken,
og vil ikke kreve justering under normale forhold.
Følsomhetsinnstilling kan justeres mellom verdiene
1–5, der 1 er minst følsom og 5 er mest følsom.

Følsomhetsmodus-skjerm i innstilling 2

Flymodus på valgt

For å bekrefte Flymodus på, trykk på
mens
LAGRE-ikonet er synlig. Verdien vil skifte til det
svarte ordet «PÅ».

298 - NO PN MN232-C4 L | Bruksanvisning

FreeStyle Comfort
For å justere følsomhetsmodus, trykk på
mens
ENDRE-ikonet er synlig. Verdien vil skifte til nesteblåe nummer.

Følsomhetsmodus, innstilling 3 valgt

For å bekrefte innstilling 3 i følsomhetsmodus,
trykk påmens Lagre-ikonet er synlig. Verdien
vil veksle til svart «3».

AUTOSAT-MODUS
FreeStyle Comfort leveres med autoSAT-modusslått på.

Hovedskjerm i autoSAT-modus

For å slå modusen av, trykk på
mens
ENDRE-ikonet er synlig. Verdien vil skifte
til det blå ordet «AV».

For å avbryte følsomhetsmodusendringen, trykk påmens AVBRYT-ikonet er synlig.

autoSAT-modus av valgt

Følsomhetsmodus, innstilling 3, bekreftetFor å bekrefte autoSAT-modus av, trykk påmens LAGRE-ikonet er synlig. Verdien vil skifte
til det svarte ordet «AV».
For å avbryte autoSAT-modus på, trykk påmens AVBRYT-ikonet er synlig.

Følsomhetsmodus, innstilling 5, (maks) bekreftet

Trykk pånår skjermen viser «Neste»-symboletfor å flytte menysiden til neste skjermalternativ.

autoSAT-modus av bekreftetFor å slå på autoSAT-modus, gjenta skrittene
ovenfor mens du veksler til PÅ.
Trykk pånår skjermen viser «Neste»-symboletfor å flytte menysiden til neste skjermalternativ.

PN MN232-C4 L | Bruksanvisning NO - 299

FreeStyle Comfort
Feilsøking
ProblemMulig årsak

Løsning

FreeStyle Comfort
viser alarm og avgir
intermitterende pip.

Se Alarmtilstandstabellen.

Se Alarmtilstandstabellen.

Oksygenkonsentratoren slårseg ikke på.

Batteriet er utladet.

Koble enheten til et uttak fra vekselstrøm eller
likestrøm.

Ikke tilkoblet ekstern strøm.
Generell feil.

Kontroller at eksterne kontakter er sikre.
Utfør en full strømsyklus ved å fjerne batteriet og
koble fra en hvilken som helst ekstern vekseleller likestrømkilde. Sett deretter batteriet inn
igjen for å fullføre strømsyklusen. Koble til
ekstern strømkilde ved behov.
Ta kontakt med utstyrsleverandøren din og bytttil en annen oksygenkilde som nødvendig.

Alle andre problemer.

Utfør en full strømsyklus ved å fjerne batteriet og
koble fra en hvilken som helst ekstern vekseleller likestrømkilde. Sett deretter batteriet inn
igjen for å fullføre strømsyklusen. Koble til
ekstern strømkilde ved behov.
Bytt til en annen oksygenkilde, som tilgjengelig,og ta kontakt med utstyrsleverandøren din.

Merk: Noen alarmer vil føre til at enheten slår seg av.
Merk: Under et totalt strømbrudd tilbakestilles alarmsystemet (tidligere alarmer slettes), og vil bare vise en alarmhvis tilstanden fortsatt eksisterer når strømmen gjenopprettes.

300 - NO PN MN232-C4 L | Bruksanvisning

FreeStyle Comfort
Reiseinformasjon

1. Slå POC på BARE ved bruk av batteristrøm når
ingen ekstern strømkilde er tilkoblet.

FreeStyle Comfort oppfyller FAAs-retningslinjer
for bruk om bord kommersielle fly. Før enhver reise(fly, tog, båt osv.) skal du:• Sikre at konsentratoren er i god driftstilstand.
• Kontakte flyselskapet ditt minst 48 timer før
avgang for å informere dem om at du reiser med
en FreeStyle Comfort.
Vi forslår at du alltid har med deg ekstra batterier
på reiser. Ved flyreise må du sørge for nok
batteri til å dekke bakketiden (før innsjekking,sikkerhetskontroll osv.) og mellomlandinger.
De fleste flyselskaper krever at du har ettilstrekkelig antall fulladede batterier til å drive
enheten i minst 150 % av den forventet maksimale
reisetiden (flytider, bakketid før og etter flyturen,
videre flyvninger og for uventede forsinkelser).

2. Aktiver flymodus med én av følgende metoder:a. Følg instruksjonene for å aktivere flymodus
i avsnittet Instruksjoner for menynavigering.

Merk: Hvis du flyr, anbefales det å bære ekstra
batterier i en sikker bærepose.

FAA-REGULERING

Ekstra batterier må være beskyttet hver for seg forå hindre kortslutning, ved å plasseres i den
originale emballasjen eller ved at polene isolerespå annen måte, f.eks. ved å tape over de eksponerte
polene eller plasser hver batteri i en egen plastpose
eller beskyttende pose, og skal bare transporteresi håndbagasjen.

b. Trykk og hold inne både opp- og nedknappene i 5 sekunder til du hører et langt pip og
ser at flysymbolet vises.

AKTIVER FLYMODUS
Batterispenning
POC kan bare brukes på et fly under normale
driftsprosedyrer, BARE ved bruk av batteristrømnår ingen ekstern strømkilde er tilkoblet.
Ekstern strøm
Dersom det finnes en stikkontakt på et fly, kan den
fungere som en reserveløsning for POC-batterierhvis den er brukbar stand på et egnet sted til å
brukes som en POC-strømkilde. I dette tilfellet må
følgende anbefalte fremgangsmåte for overgang fra
batteri til elektrisk strøm i fly følges:

3. Plugg inn POC når flysymbolet vises.

Merk: På grunn av flybegrensninger lades ikke
batteriene opp i flymodus.

PN MN232-C4 L | Bruksanvisning NO - 301

FreeStyle Comfort
Tilbehør

PLASSERE ENHETEN I BÆREVESKEN

For riktig ytelse og sikkerhet skal du bare bruke
tilbehøret i listen som er levert av CAIRE gjennom
din oksygenleverandør. Bruk av tilbehør som duikke finner i listen nedenfor, kan svekke ytelsen
og/eller sikkerheten til konsentratoren.

1. Sett enheten i bunnen av vesken, med filtreog batteriåpninger riktig innrettet.

Standard tilbehør for FreeStyle ComfortBæreveske

MI396-1

Ryggsekk

MI459-1

Skulderstropp

MI397-1

Strømforsyning for likestrøm(alle ledninger inkludert)

PW036-1S

Strømledning til likestrøm

CD041-1

Eksternt grovpartikkelfilter (høyre)FI226-1

Eksternt grovpartikkelfilter (venstre)

FI227-1

2. Fest glidelåsen rundt bunnen av posen.

Tilgjengelig valgfritt tilbehør
Ytterligere batteripakke med 8 celler

BT034-1S

Batteripakke med 16 celler

BT035-1S

Bordlader for batteripakke

BT036-1

Tilbehørsveske

MI372-2

Merk: Flere alternativer kan være tilgjengelige forlandspesifikke strømledninger som notert ovenfor.
Ta kontakt med CAIRE eller din oksygenleverandør hvis
alternative muligheter er nødvendige for bestillingen.

MONTERING AV SKULDERSTROPP
1. Skulderstropp kan festes enten til enheten direkteeller til bæreposen.
2. Klem stroppen på ringen på enheten eller vesken.

Instruksjoner for bæreveske
Bruk følgende instruksjoner for å plassere og feste
fast din FreeStyle Comfort-enhet i bærevesken.

302 - NO PN MN232-C4 L | Bruksanvisning

FreeStyle Comfort
Kassering og gjenbruk
Returner alltid FreeStyle Comfort, med allekomponenter, til utstyrsleverandøren din for
riktig kassering. Du kan også kontakte lokale
myndigheter for anvisning om korrekt deponering
av batteriet. Enheten kan gjenbrukes mellom
personer, men må returneres til utstyrsleverandøren
for reprosessering.

WEEE og RoHS
Dette symbolet skal påminne eierne av
utstyret om at utstyret skal leveres til et gjenvinningsanlegg når brukstiden er omme, i henhold til
direktiv om avfall fra elektrisk og elektronisk utstyr(WEEE-direktivet).

Materialer i direkte kontakt med
operatøren• Konsentratorens deksel: PC/ABS• Konsentratorens kontrollpanel: Polyester EBG7
eller tilsvarende• Oksygenutløp, dyse: Aluminium• Enhetens etikett: Lexan• Ledningskontakter: Polykarbonat/vinylklorid• Strømledning: PVC, metall• Strømforsyning: Lexan 940 (polykarbonat)• Batteripakke: PC/ABS• Batteripakke, strømforsyningsetiketter:
Polyesterfilm• Bæreveske til konsentratoren: Cordura, neopren,nylonforing• Stropp: Polypropylen, acetyl, vinyl, Cordura,
neopren

Produktene våre skal være i samsvar med begrensning av farlige stoffer (RoHS-direktivet). De vil
ikke inneholde mer enn spormengder av bly eller
andre farlige materialer.
FORSIKTIG: For riktig deponering
av batteriene, ta kontakt med utstyrsleverandøren eller lokale myndigheter
for krav til deponering.
FORSIKTIG: Kast alltid molekylsikt i samsvar
med alle føderale, statlige og lokale
miljøkontrollforskrifter.

PN MN232-C4 L | Bruksanvisning NO - 303

FreeStyle Comfort
EMC-testing
Veiledning og produsentens erklæring – elektromagnetisk stråling
FreeStyle Comfort er ment brukt i det elektromagnetiske miljøet som er spesifisert nedenfor. Kunden eller brukeren
av FreeStyle Comfort skal påse at det brukes i et slikt miljø.
Utslippstest
RF-utslipp
CISPR 11
RF-utslipp
CISPR 11
Harmoniske emisjoner
IEC 61000-3-2
Spenningssvingninger/
flimmerutslipp
IEC 61000-3-3

SamsvarElektromagnetisk miljø – veiledning

Gruppe 1

FreeStyle Comfort bruker bare RF-energi til sin indre funksjon.
Derfor er RF-utslippet meget lavt, og det er ikke sannsynlig
at det fører til forstyrrelser for elektronisk utstyr i nærheten.

Klasse B
Klasse A
Samsvarer

304 - NO PN MN232-C4 L | Bruksanvisning

FreeStyle Comfort passer for bruk i alle typer bygninger, inklusive
bosteder og de som er direkte koblet til offentlig lavspent
forsyningsnettverk, som gir strøm til bygninger som brukes
for boformål.

FreeStyle Comfort
Veiledning og produsentens erklæring – elektromagnetisk immunitet

FreeStyle Comfort er beregnet til å brukes i det elektromagnetiske miljøet som er spesifisert nedenfor. Kunden eller
brukeren av FreeStyle Comfort skal påse at det brukes i et slikt miljø.
Immunitetstest
IEC 60601 testnivå
Samsvarsnivå
Elektromagnetisk miljø – veiledning
Gulvet bør være av tre, betong eller
Elektromagnetisk±8 kV-kontakt±8 kV-kontaktkeramiske fliser. Hvis gulvene er dekket
miljø – veiledning
med syntetiske materialer, skal den±15 kV luftutslipp±15 kV luftutslipp
IEC 61000-4-2
relative fuktigheten være minst 30 %.
±2 kV for strømforsy±2 kV for strømforsyningslinjer
Elektrisk rask
Kvaliteten på nettstrømmen skal tilsvare
ningslinjer
forbigående/burst
kvaliteten i en/et typisk helseinstitusjon
eller hjemmepleiemiljø.
IEC 61000-4-4±1 kV for inndata/
Ikke aktuelt
utdata-linjer±1 kV linje(r) til linje(r)Overspenning
IEC 61000-4-5±2 kV fellesmodus
på vekselstrømlinjer±1 kV linje(r) til linje(r)±1 kV differensial påvekselstrømlinjer

Kvaliteten på nettstrømmen skal tilsvare
kvaliteten i en/et typisk helseinstitusjon
eller hjemmepleiemiljø.

±2 kV fellesmodus på
utendørs I/O-linjer
0 % UT for 0,5 syklus
0 % UT i 0,5 syklus(0°, 45°, 90°, 135°,(0°, 45°, 90°, 135°,
180°, 225°, 270°, 315°) 180°, 225°, 270°, 315°) Kvaliteten på nettstrømmen skal tilsvare
kvaliteten i en/et typisk helseinstitusjon
Spenningsfall, korteeller hjemmepleiemiljø. Hvis brukeren av
0 % UT for 1 syklus (0°) 0 % UT i 1 syklus (0°)
avbrudd og spenFreeStyle Comfort trenger kontinuerlig
ningsvariasjoner
drift under et strømbrudd, anbefales
i strømforsyningen 70 % U (30 % fall i U ) 70 % U (30 % fall i U )det at FreeStyle Comfort drives av
T
T
T
T
IEC 61000-4-11
for 25/30 sykluser (0°)
i 25/30 sykluser (0°)
en avbruddsfri strømforsyning eller etbatteri.
0 % UT for 250/300
sykluser (0°)

0 % UT i 250/300
sykluser (0°)

Strømfrekvens(50/60 Hz)
3 A/m
3 A/m
magnetfelt
IEC 61000-4-8
MERK: UT er vekselstrømsnettspenningen før anvendelse av testnivået.

Strømfrekvensmagnetfeltene børvære på nivå med en typisk plassering
i en/et typisk helseinstitusjon eller
hjemmepleiemiljø.

MERK: I noen tilfeller kan Freestyle Comfort bli tilbakestilt etter en betydelig elektrostatisk utladning, og kan kreve
en full strømsyklus for å gjenoppta normal drift. Se avsnittet Feilsøking for flere detaljer.

PN MN232-C4 L | Bruksanvisning NO - 305

FreeStyle Comfort
Veiledning og produsentens erklæring – elektromagnetisk immunitet

FreeStyle Comfort er ment brukt i det elektromagnetiske miljøet som er spesifisert nedenfor. Kunden eller brukeren
av FreeStyle Comfort skal påse at det brukes i et slikt miljø.
Immunitetstest IEC 60601 testnivå
Samsvarsnivå
Elektromagnetisk miljø – veiledning
10 VrmsBærbart og mobilt RF10 Vrms
Ledet RFkommunikasjonsutstyr skal ikke brukes
150 kHz til 80 MHz
IEC 61000-4-6nærmere noen del av FreeStyle Comfort,
inkludert kabler, enn den anbefalte
avstanden beregnet ut fra ligningen som
gjelder for senderfrekvensen.
10 V/m
10 V/mUtstrålt RF
IEC 61000-4-3

80 MHz til 2700 MHz

Anbefalt avstandd = 1.2 P
d = 1.2 P = 80 MHz til 800 MHz
d = 1.2 P = 800 MHz til 2700 MHz
der P er maksimal utgangseffekt
for senderen i watt (W) i henhold til
produsenten av senderen, og d er den
anbefalte avstanden i meter (m).
Feltstyrker fra faste RF-sendere,som bestemt av en elektromagnetisk
undersøkelse,a skal være mindre enn
samsvarsnivået i hvert frekvensområde. b
Det kan forekomme forstyrrelseri nærheten av utstyr merket med
følgende symbol:

MERK 1 Ved 80 MHz og 800 MHz gjelder høyere frekvensområdet.
MERK 2 Retningslinjene gjelder kanskje ikke i alle situasjoner. Elektromagnetisk spredning påvirkes av absorpsjonog refleksjon fra strukturer, gjenstander og mennesker.
aFeltstyrker fra faste sendere, så som basestasjoner for radiotelefoner (mobil/trådløs) og walkie-talkies,
amatørradioer, AM og FM radio- og TV-sending kan ikke forutsies teoretisk med nøyaktighet. For å vurdere det
elektromagnetiske miljøet ved faste RF-sendere, bør en elektromagnetisk stedsundersøkelse vurderes. Hvis
den målte feltstyrken på stedet hvor FreeStyle Comfort brukes overskrider gjeldende RF-samsvarsnivå ovenfor,bør FreeStyle Comfort observeres for å bekrefte normal drift. Hvis unormal ytelse observeres, kan ytterligere tiltak
være nødvendig, slik som reorientering eller omplassering av FreeStyle Comfort.
bOver frekvensområdet mellom 150 kHz og 80 MHz skal feltstyrkene være lavere enn 10 V/m.

306 - NO PN MN232-C4 L | Bruksanvisning

FreeStyle Comfort
Anbefalt avstand mellom bærbart og mobilt RF-kommunikasjonsutstyr
og FreeStyle Comfort

FreeStyle Comfort er egnet for bruk i et elektromagnetisk miljø hvor utstrålte RF-forstyrrelser er kontrollert. Kundeneller brukeren av FreeStyle Comfort kan bidra til å forhindre elektromagnetisk forstyrrelse ved å opprettholdeen minimumsavstand mellom bærbart og mobilt RF-kommunikasjonsutstyr (sendere) og FreeStyle Comfort,
som anbefalt nedenfor, i henhold til maksimal utgangseffekt for kommunikasjonsutstyr.
Maksimal nominell
Avstand i henhold til senderens frekvens
utgangseffekt formsenderen
150 kHz til 80 MHz
80 MHz til 800 MHz
800 MHz til 2,5 GHz

d = 1.2 P
d = 1.2 P
d = 1.2 P
W
0,01
0.12
0.12
0.23
0,1
0.38
0.38
0.73
1
1.2
1.2
2.3
10
3.8
3.8
7.3
100
12
12
23
For sendere som er klassifisert med en maksimal utgangseffekt som ikke er oppført ovenfor, kan den anbefalte
avstanden d i meter (m) beregnes ved hjelp av ligningen som gjelder for senderfrekvensen, der P er den maksimale
utgangseffekten for senderen i watt (W) ifølge produsenten av senderen.
MERK 1 Ved 80 MHz og 800 MHz gjelder avstanden for høyere frekvensområder.
MERK 2 Retningslinjene gjelder kanskje ikke i alle situasjoner. Elektromagnetisk spredning påvirkes av absorpsjonog refleksjon fra strukturer, gjenstander og mennesker.

Klassifisering

Type beskyttelse mot elektrisk støt:
Klasse II	Beskyttelse mot elektrisk støt			oppnås ved dobbel isolasjon.
Grad av beskyttelse mot elektrisk støt:	Type BF Utstyr som gir en bestemt
		grad av beskyttelse mot elektrisk sjokk vedr
		1) tillatt lekkasjestrøm,		2) pålitelighet av beskyttende jordforbindelse		(hvis til stede).
		Ikke tiltenkt direkte kardial anvendelse.
Uavhengig testing for medisinsk elektrisk
utstyrsstandard:	Testet til å overholde• IEC 60601-1 utgave 3.1: Medisinsk elektrisk
utstyr – del 1: Generelle krav for sikkerhet.
• CAN/CSA C22.2 nr. 60601-1-8 utgave 2,1,
Medisinsk elektrisk utstyr – del 1: Generelle
sikkerhetskrav.
Beskyttelse mot potensiell elektromagnetisk eller
annen interferens mellom utstyret og andre enheter.
	• Testet for å være i samsvar med EN 60601-1-2/IEC 60601-1-2: 4. utgave.

	• Testet til å overholde RTCA/DO160
	Avsnitt 21 kategori M.
• CISPR 11 / EN 55011 Klasse B gruppe 1,	«Industriell, vitenskapelig og medisinsk (ISM)	utstyr».
• FCC Del 15, underdel B – klasse B	Utilsiktede radiatorer.
IP22 – Beskyttelse mot inntrengning av fingre
eller lignende gjenstander større enn 12,5 mm
diameter. Vertikalt dryppende vann skal ikke haskadelig effekt når kabinettet er tiltet i en vinkel
på 15° fra sin normale posisjon.
Tillatt metode for rengjøring og infeksjonskontroll:
Se avsnittet «Rengjøring, stell og vedlikehold»i denne bruksanvisningen for FreeStyle.
Grad av sikkerhet ved bruk i nærvær av brennbare
anestesigasser:	Utstyr ikke tilpasset slik bruk.
Driftsmodus: Kontinuerlig drift.

PN MN232-C4 L | Bruksanvisning NO - 307

FreeStyle Comfort
Användarkontroller och indikatorer för systemstatus
ISO 7000
Bruksanvisning; driftsanvisningar.
Reg.nr 1641
Land och tillverkningsdatum.
”CC” identifierar tvåbokstavskoden
för tillverkningslandet.
Tillverkningsdatumet är i formatetÅÅÅÅ-MM-DD. Reg.nr 6049

ISO 7010: Grafiska symboler – säkerhetsfärger
och säkerhetssymboler – registreradesäkerhetssymboler
Bruksanvisningen måste läsas.
Reg.nr M002Håll på avstånd från öppna lågor, eld
och gnistor. Öppna antändningskällor
och rökning förbjuden. Reg.nr P003

Skydda den mot regn, håll den torr.
Reg.nr 0626

Rök inte nära enheten eller när
du använder den. Reg.nr P002

Staplingsgräns per nummer, där ”n”
anger det maximala antalet tillåtna
enheter. Reg.nr 2403
Katalognummer. Reg.nr 2493

Varning. Reg.nr W001
IEC 60417
Patientansluten del av typ BF (grad av
skydd mot elektrisk stöt). Reg.nr 5333

Serienummer. Reg.nr 2498
Temperaturintervall för förvaring eller
användning. Reg.nr 0632
Luftfuktighetsintervall för förvaring.
Reg.nr 2620
Begränsning av atmosfärtryck.
Reg.nr 2621
Tillverkarens namn och adress.
Reg.nr 3082

Klass II-utrustning, dubbelisolerad.
Reg.nr 5172
EU-förordning 2017/745
Auktoriserad representant i den
Europeiska gemenskapen
Om produktens unika enhetsidentifierare(UDI) har symbolen CE#### uppfyller
den kraven för medicintekniska
produkter. Symbolen CE#### anger
numret för anmält organ.
Ytterligare symboler

Denna sida upp. Reg.nr 0623

Håll på avstånd från lättantändliga
material, olja och fett.

Ömtåligt, hanteras varsamt.
Reg.nr 0621

Får inte demonteras.
Ökar flödet.

Innehåller farliga ämnen. Reg.nr 3723

Minskar flödet.

Importör. Reg.nr 3725

Menyknapp.

O2-utgång

Strömbrytare. Reg.nr 5009
RTCA/DO160 avsnitt
21 kategori
M-kompatibel.

308 - SV PN MN232-C4 L | Bruksanvisning

Tillverkaren av denna POC har fastställt
att enheten uppfyller alla tillämpliga
FAA-kriterier för godkännande ochanvändning ombord på flygplan.

FreeStyle Comfort
Tillverkaren av denna POC har fastställt
att enheten uppfyller alla tillämpliga
FAA-kriterier för godkännande ochanvändning ombord på flygplan.

EU-rådets direktiv 2012/19/EU
WEEE
Syftet med denna symbol är att
påminna ägaren av utrustningen
att lämna tillbaka den till enåtervinningsanläggning när den ska
kasseras, enligt WEEE-direktivet om
kassering av elektriskt avfall (Waste
Electrical and Electronic Equipment).
Våra produkter uppfyller kraven
i RoHS-direktivet (begränsning
av användningen av vissa farligaämnen i elektriska och elektroniska
produkter). De innehåller inte mer änspårelement av bly eller andra farliga
material.

Återvinningssymbol
Auktoriserad representant i Schweiz(ADD)
Om enheten är försedd med
UKCA-märket som visas med
UKCA#### som anger det anmälda
organnumret uppfyller denna enhet
UKCA-föreskrifter.
21 CFR 801.15
Enligt amerikansk federal lag får
denna anordning endast säljas av
läkare eller på läkares anmodan.

IEC 60601-1-8
Tekniskt larm med låg prioritet

ETL-certifiering
Certifierad för både amerikanska
och kanadensiska marknader
enligt gällande amerikanska ochkanadensiska standarder
Innehåller FCC-ID: WAP2001
FCC-meddelande
Denna enhet kan innehålla CYBLE-022001-00,
inklusive antennen 2450AT18B100 från Johnson
Technology, och överensstämmer med del15 av FCC-reglerna. Enheten uppfyller kraven
för godkännande av modulära sändare enligt
FCC:s offentliga meddelande DA00-1407.
Användning av sändaren gäller enligt följande
villkor: 1) Enheten får inte orsaka skadliga
störningar och 2) Enheten måste klara eventuellastörningar, inklusive störningar som kan orsaka
oönskad drift. Produkten kan omfattas av ett
eller flera amerikanska eller internationella
patent. Besök vår webbplats nedan för att
se en lista över tillämpliga patent. Pat:www.caireinc.com/corporate/patents

ISO 15223-1
Medicinteknisk produkt. Reg.nr 5.7.7
Unikt enhetsidentifieringsnummer
5.7.10A S/NZS 4417.1, AS/NZS 4417.2
ACMA Regulatory ComplianceMark – indikerar efterlevnad av
ACMA:s regelverk (ADD)Trådlös teknik
Anslutningstyper

SSP, iAP2, GATT

Frekvens

2402 to 2480 MHz

Max RF-uteffekt+4 dBmArbetsområde

10 m (klass 2)

Vi rekommenderar att maskinen befinner sig på minst
1,1 cm avstånd från kroppen under drift. Gäller intemasker, luftslangar eller tillbehör.

Grafiska symboler för användning på utrustning
IEC 60417
Direktström. Reg.nr 5031
Växelström. Reg.nr 5032
IEC 60601-1
Droppskyddad

PN MN232-C4 L | Bruksanvisning SV- 309

FreeStyle Comfort
Packa upp FreeStyle Comfort
Kontrollera att alla
komponenter som listas finns
med i paketet. Om något
saknas ska du omedelbart
kontakta leverantören av
syrgasutrustningen.

Standardtillbehör• Bärbar syrgaskoncentrator• Transportväska• Axelrem• 8-cellsbatteri• Nätaggregat• Strömkabel• Likströmsadapter• Användarhandbok

Extra tillbehör• Extra 8-cellsbatteri• 16-cells batteripaket• Bordsladdare• Tillbehörsväska

Bekanta dig med FreeStyle Comfort
Bekanta dig med de viktigaste FreeStyle Comfort-funktionerna och dess användargränssnitt.

Framsida
RemfästeAnvändargränssnitt

Syrgasutlopp

Luftutlopp

Luftintag/filter
Luftintag/filter

Strömuttagsflik

Batteripaket
Frigöringsknapp
för batteri
310 - SV PN MN232-C4 L | Bruksanvisning

FreeStyle Comfort
Ovansida
VisaÖkar flödet

Strömknapp

Menyknapp

Minskar flödet

Display för batteri
Inställning av flöde

Batteridisplay

autoSAT-indikatorÅterstående
batteriprocent

Larmdisplay
Indikator för
bolusdosering

Display för växelström/likström

Inställning
av flöde

Displayplats
för larm

Batteridisplay

Laddningsindikator

PN MN232-C4 L | Bruksanvisning SV- 311

FreeStyle Comfort
Vad är en syrgaskoncentrator?
Luften vi andas innehåller cirka 21 % syre,
78 % kväve och 1 % andra gaser. I FreeStyle®
Comfort® sugs rumsluft in i maskinen genom
luftintagen. Luften passerar därefter genom ett
adsorberande material som kallas molekylsikt.
Detta material separerar syret från kvävet och
släpper endast igenom syret. Resultat är ett flöde
av högkoncentrerad syrgas som levereras till
användaren.
I FreeStyle Comfort kombineras avancerade
tekniker för koncentration och konservering
av syrgas för att skapa en portabel enhet som
ger maximal bärbarhet och rörelsefrihet.
Den avancerade pulsflödestillförseln känner snabbt
av när användaren andas in och ger en syrgaspuls
i början av varje inandning. Pulsflödestillförsel
är idealiskt vid aktiviteter och vistelser utanför
hemmet. Det sparar syrgas och gör att batteriet
räcker längre än vid ett kontinuerligt flöde.
FreeStyle Comfort är tillräckligt lätt för att bäras
i en axelrem. Den kan drivas med växelström(vägguttag), likström (i fordon) eller batteri.

Avsedd användning
CAIRE FreeStyle Comfort syrgaskoncentrator
är avsedd för administrering av extra syre. Enheten
är inte avsedd för livsuppehållande verksamhet ochden är inte ämnad för patientövervakning.

Varför har din läkare ordinerat
syrgas?
Många människor lider av olika hjärt- ochlungsjukdomar samt andra andningssjukdomar.
Ett betydande antal av dessa kan dra nytta av
kompletterande syrgasbehandling hemma, på resor
eller när de deltar i dagliga aktiviteter utanför
hemmet.
Syrgas är en gas som utgör 21 % av luften vi
andas och våra kroppar är beroende av en konstanttillförsel för att fungera ordentligt. Din läkare
har ordinerat ett flöde eller en inställning som
behandling för just dina andningsbesvär.

312 - SV PN MN232-C4 L | BruksanvisningÄven om syrgas inte är en beroendeframkallandedrog kan en ej godkänd syrgasbehandling varafarlig. Du måste rådgöra med en läkare innan
du använder syrgaskoncentratorn. Leverantören
av syrgasutrustningen visar hur den ordinerade
flödesfrekvensen ställs in.

Användarprofil
Koncentratorer används för att ge kompletterande
syrgas till användare som upplever obehag pågrund av sjukdomar som påverkar lungornas
effektivitet när det gäller att överföra syre i luften
till blodomloppet. Bärbara syrgaskoncentratorer
används inte för att lagra eller förvara syre.
De behöver inte fyllas på och kan laddas överallt
där växelström eller likström är tillgänglig(förutom på ett flygplan). Det gör att användaren
blir förhållandevis oberoende och kan använda
utrustningen i hemmet, på rörlig fot (både i ochutanför hemmet), på resor och vid olika aktiviteter.
Syrgaskoncentratorn får endast användas på
ordination av läkare.
Den typiska syrgasanvändaren är äldre än 65 år
och lider av olika luftvägssjukdomar, t.ex. kronisk
obstruktiv lungsjukdom (KOL). Användarna har
vanligtvis god kognitiv förmåga och måste kunna
meddela om obehag uppstår. Om användaren inte
kan meddela upplevt obehag, eller inte kan läsa och
förstå koncentratorns märkning eller bruksanvisning,bör användning endast ske under överinseende avnågon som kan göra detta. Vid obehag i samband
med användning av koncentratorn bör användaren
kontakta vårdgivaren. Användaren bör ävenha tillgång till reservsyrgas (dvs. en syrgastub)om det blir strömavbrott eller koncentratorn
slutar att fungera. Användaren behöver ingensärskild kunskap eller förmåga för att använda
koncentratorn.

FreeStyle Comfort
Viktigt!
Säkerhetsinstruktioner definieras enligt följande:
VARNING: VIKTIG SÄKERHETSINFORMATION FÖR FAROR SOM KAN ORSAKA ALLVARLIG PERSONSKADA.
VAR FÖRSIKTIG: Viktig information för att
förhindra skador på FreeStyle Comfort.
Obs! Särskild uppmärksamhet bör ägnas
åt informationen.

Indikationer för användning
VARNING: DET ÄR MYCKET VIKTIGTATT ENBART STÄLLA IN DEN ORDINERADE NIVÅN AV SYRE. ÄNDRA INTEFLÖDESVALET OM DU INTE HAR BLIVIT
UPPMANAD ATT GÖRA DET AV EN
KVALIFICERAD KLINIKER.
VARNING: DENNA ENHET HAR INTE TESTATSELLER GODKÄNTS FÖR ANVÄNDNING PÅ BARN.
VARNING: ENLIGT FEDERAL LAGSTIFTNING I
USA FÅR DENNA ENHET ENDAST SÄLJAS ELLERHYRAS UT PÅ LÄKARES ELLER ANNAN UTBILDADSJUKVÅRDSPERSONALS ORDINATION.
VARNING: ENHETEN ÄR INTE AVSEDD FÖRLIVSUPPEHÅLLANDE BRUK. YTTERLIGAREÖVERVAKNING KAN KRÄVAS FÖR ANVÄNDARE
SOM INTE KAN KOMMUNICERA OBEHAG VIDANVÄNDNING AV ENHETEN. ANVÄNDARE MED
HÖRSEL- OCH/ELLER SYNNEDSÄTTNINGAR KAN
BEHÖVA HJÄLP MED ATT ÖVERVAKA LARM.
VARNING: INSTÄLLNINGAR FÖR PULSDOSLÄGE
MÅSTE BESTÄMMAS INDIVIDUELLT BASERATPÅ VARJE ANVÄNDARES BEHOV I VILA, UNDERTRÄNING ELLER VID RESA.
VARNING: DENNA ENHET ÄR AVSEDDFÖR ANVÄNDNING INOM HEMSJUKVÅRD OCH
LIKARTADE ÖPPENVÅRDSMILJÖER.

VARNING: VISSA ANDNINGSANSTRÄNGNINGARFRÅN PATIENTEN KANSKE INTE UTLÖSER
KONSERVERINGSFUNKTIONEN.
VARNING: FREESTYLE COMFORTÄR INTE AVSEDD ATT ANVÄNDAS AV EN
TRAKEOSTOMERAD PATIENT.
VARNING: GRAVIDA ELLER AMMANDEKVINNOR SKA INTE ANVÄNDA TILLBEHÖRENSOM REKOMMENDERAS I DENNA HANDBOKEFTERSOM DE KAN INNEHÅLLA FTALATER.

Kontraindikationer vid användning
VARNING: UNDER VISSA
OMSTÄNDIGHETER KAN
ANVÄNDNINGEN AV ICKE-ORDINERATSYRE VARA FARLIG. DEN HÄR
ENHETEN BÖR ENDAST ANVÄNDAS
VID ORDINATION FRÅN EN LÄKARE.
VARNING: DENNA ENHET ÄR INTE AVSEDD
ATT ANVÄNDAS I ELLER I NÄRHETEN AV
UTRYMMEN MED MAGNETISK RESONANS,
T.EX. MR-APPARATER.
VARNING: FÅR INTE ANVÄNDAS I NÄRHETEN AVLÄTTANTÄNDLIGA ANESTETIKA.
VARNING: I LIKHET MED ALLA ELDRIVNA
ENHETER KAN DRIFTEN TILLFÄLLIGT AVBRYTASTILL FÖLJD AV STRÖMAVBROTT ELLER PÅ
GRUND AV ATT SYRGASKONCENTRATORNBEHÖVER SERVAS AV EN BEHÖRIG TEKNIKER.
SYRGASKONCENTRATORN ÄR INTE LÄMPLIGFÖR ANVÄNDARE VARS HÄLSA DRABBAS AV
ALLVARLIGA KONSEKVENSER VID SÅDANA
TILLFÄLLIGA AVBROTT.
VARNING: DENNA ENHET ÄR INTE GODKÄNDFÖR ANVÄNDNING I ALLA TYPER AV MILJÖER.
EXPONERING FÖR VISSA ELEKTROMAGNETISKAKÄLLOR KAN LEDA TILL STÖRNINGAR
ELLER OÖNSKAD FUNKTION HOS ENHETEN.
VID STÖRNINGAR SKA ENHETEN OMEDELBARTSTÄNGAS AV.

VARNING: DENNA ENHET ÄR AVSEDD
ATT ANVÄNDAS AV PATIENTER SOM FÅRSYRGASBEHANDLING.

PN MN232-C4 L | Bruksanvisning SV- 313

FreeStyle Comfort
Säkerhetsanvisningar
VARNING: DET ÄR INTE TILLÅTET ATT
MODIFIERA DENNA UTRUSTNINGVARNING: DENNA PRODUKT LEVERERAR SYRE AV
HÖG KONCENTRATION SOM UNDERLÄTTAR SNABBFÖRBRÄNNING. TILLÅT INTE RÖKNING ELLERÖPPEN ELD I SAMMA RUM SOM 1) DENNA ENHETELLER 2) SYREFYLLDA TILLBEHÖR. OM MAN INTE
TAR HÄNSYN TILL DENNA VARNING KAN DET
LEDA TILL ALLVARLIG BRAND, EGENDOMSSKADA,KROPPSSKADA OCH/ELLER DÖD.
VARNING: ANVÄND INTE ENHETEN I ETTBEGRÄNSAT UTRYMME (T.EX. ETT LITET SKÅPELLER EN VÄSKA) DÄR VENTILATIONEN ÄRBEGRÄNSAD. DETTA KAN LEDA TILL ATTSYRGASKONCENTRATORN ÖVERHETTAS OCH
ATT PRESTANDAN FÖRSÄMRAS.
VARNING: KONCENTRATORN BÖR PLACERASSÅ ATT DEN INTE UTSÄTTS FÖR RÖK,FÖRORENINGAR ELLER ÅNGOR.
VARNING: OM DET SKULLE INTRÄFFA ETTALLVARLIGT TILLBUD MED DEN HÄR ENHETENBÖR ANVÄNDAREN OMEDELBART RAPPORTERA
HÄNDELSEN TILL LEVERANTÖREN OCH/ELLERTILLVERKAREN. MED ALLVARLIGT TILLBUD AVSESSKADA, DÖDSFALL ELLER POTENTIAL ATT ORSAKA SKADA/DÖDSFALL OM DET SKULLE INTRÄFFA
IGEN. ANVÄNDAREN KAN OCKSÅ RAPPORTERA
HÄNDELSEN TILL DEN BEHÖRIGA MYNDIGHETEN I
DET LAND DÄR HÄNDELSEN INTRÄFFADE.
VARNING: ANVÄNDNING AV VISSATILLBEHÖR FÖR SYRGASHANTERING
SOM INTE ÄR DIREKT AVSEDDAATT ANVÄNDAS MED DENNASYRGASKONCENTRATOR KAN
PÅVERKA DESS PRESTANDANEGATIVT. DENNA BRUKSANVISNINGBESKRIVER VILKA TILLBEHÖR SOM
REKOMMENDERAS.
VARNING: VAR VÄNLIG KONTAKTAUTRUSTNINGSLEVERANTÖREN FÖRINSPEKTION OCH MÖJLIG REPARATION AV
SYRGASKONCENTRATORN OM ENHETEN HAR
TAPPATS, SKADATS ELLER EXPONERATS FÖRVATTEN. ANVÄND INTE SYRGASKONCENTRATORNOM NÄTSLADDEN ELLER KONTAKTEN ÄR SKADAD.
VARNING: FÄST SYRGASKONCENTRATORN
ORDENTLIGT, MED BÄLTE ELLER PÅ ANNAT
SÄTT, NÄR DEN TRANSPORTERAS I ETT FORDONFÖR ATT UNDVIKA SKADOR.
314 - SV PN MN232-C4 L | Bruksanvisning

VARNING: U.S. DEPARTMENT OF
TRANSPORTATIONS (DOT) OCH FÖRENTA
NATIONERNAS (FN) FÖRORDNINGAR KRÄVER
ATT BATTERIET AVLÄGSNAS FRÅN ENHETENFÖR ALLA INTERNATIONELLA FLYGRESORDÄR SYRGASKONCENTRATORN CHECKASIN SOM BAGAGE. BATTERIET MÅSTE OCKSÅ
AVLÄGSNAS FRÅN ENHETEN OCH PAKETERASORDENTLIGT VID FÖRSÄNDELSE AVSYRGASKONCENTRATORN.
VARNING: ANVÄND INTE DENNA UTRUSTNINGUTAN ATT FÖRST LÄSA OCH FÖRSTÅDENNA BRUKSANVISNING. KONTAKTA
ÅTERFÖRSÄLJAREN INNAN DU FÖRSÖKERANVÄNDA UTRUSTNINGEN OM DU INTEFÖRSTÅR VARNINGARNA OCH ANVISNINGARNA,
ANNARS KAN FÖLJDEN BLI PERSON- ELLER
MATERIALSKADOR.
VARNING: OM DU KÄNNER OBEHAG ELLER
UPPLEVER EN MEDICINSK NÖDSITUATION SKADU SÖKA LÄKARHJÄLP OMEDELBART.
VARNING: ANVÄNDNING AV SYRGASKONCENTRATORN UTANFÖR DET SPECIFICERADEDRIFTSTEMPERATURINTERVALLET KAN
BEGRÄNSA KONCENTRATORNS MÖJLIGHET ATTUPPFYLLA SPECIFIKATIONEN FÖR SYRGASKONCENTRATION. SE SPECIFIKATIONSAVSNITTET IDENNA BRUKSANVISNING FÖR INFORMATION OM
TEMPERATURGRÄNSERNA.
VARNING: ANVÄND INTE OLJE-, FETT-,PETROLEUMBASERADE ELLER ANDRALÄTTANTÄNDLIGA PRODUKTER TILLSAMMANS
MED SYREFYLLDA TILLBEHÖR ELLERSYRGASKONCENTRATORN. ANVÄND ENDASTVATTENBASERADE, SYREKOMPATIBLALOTIONER OCH SALVOR. SYRE PÅSKYNDARFÖRBRÄNNINGEN AV LÄTTANTÄNDLIGA ÄMNEN.
VARNING: FELAKTIG ANVÄNDNING AV
BATTERIET KAN LEDA TILL ATT BATTERIETBLIR VARMT ELLER FATTAR ELD, VILKET KANORSAKA ALLVARLIG SKADA. SE TILL ATT INTEGÖRA HÅL I, SLÅ PÅ, TRAMPA PÅ ELLER TAPPABATTERIET, ELLER PÅ ANNAT SÄTT UTSÄTTABATTERIET FÖR KRAFTIGA STÖTAR ELLER
SLAG. ANVÄNDNING AV ETT SKADAT BATTERIKAN ORSAKA PERSONSKADOR.
VARNING: KORTSLUT INTE BATTERIETS
METALLKONTAKTER MED METALLOBJEKT,
T.EX. NYCKLAR ELLER MYNT. DETTA KAN SKAPAGNISTOR ELLER ÖVERDRIVEN VÄRME.

FreeStyle Comfort
VARNING: VID ANVÄNDNING AV DEN BÄRBARASYRGASKONCENTRATORN UTOMHUS MED
VÄXELSTRÖMSADAPTERN MÅSTE ADAPTERNANSLUTAS TILL ETT JORDFELSBRYTARSKYDDAT(GFI) UTTAG.
VARNING: SYRGASKONCENTRATORN BÖR INTEANVÄNDAS DIREKT INTILL ELLER OVANPÅ
ANNAN UTRUSTNING. OM DETTA INTE GÅR ATTUNDVIKA BÖR ENHETEN BEVAKAS FÖR ATTSÄKERSTÄLLA NORMAL DRIFT.
VARNING: ANVÄNDNING AV ANDRA KABLAR
OCH ADAPTRAR ÄN DE SPECIFICERADE,FÖRUTOM SÅDANA SOM TILLVERKAREN AV
DEN MEDICINSKA UTRUSTNINGEN SÄLJER
SOM ERSÄTTNINGSDELAR FÖR INGÅENDE
KOMPONENTER, KAN ORSAKA ÖKADEMISSION ELLER FÖRSÄMRAD IMMUNITET HOSSYRGASKONCENTRATORN.
VARNING: ANVÄND ENDAST DEN
ELEKTRISKA SPÄNNINGEN SOM ANGES PÅ
SPECIFIKATIONSETIKETTEN PÅ ENHETEN.
VARNING: ANVÄND INTE FÖRLÄNGNINGSSLADDARMED DENNA ENHET OCH UNDVIK ATT ANSLUTAFÖR MÅNGA STICKPROPPAR I SAMMA ELUTTAG.
ANVÄNDNING AV FÖRLÄNGNINGSSLADDAR KAN
PÅVERKA PRODUKTENS PRESTANDA NEGATIVT.
FÖR MÅNGA STICKPROPPAR I ETT ELUTTAG KAN
LEDA TILL EN ÖVERBELASTNING I ELCENTRALENVILKET GÖR ATT PROPPEN/SÄKRINGEN
LÖSES UT ELLER BRAND OM PROPPEN ELLERSÄKRINGEN INTE FUNGERAR.
VAR FÖRSIKTIG: Läs avsnittet
Felsökning i den här bruksanvisningenom ett larm utlöses eller om du upptäcker
att syrgaskoncentratorn inte fungerar
som den ska. Kontakta återförsäljaren om
du inte kan lösa problemet.
VAR FÖRSIKTIG: Följ alla anvisningar fråntillverkaren för att undvika att garantin upphävs.

VARNING: UTFÖR INGETUNDERHÅLLSARBETE ANNAT ÄN DE
MÖJLIGA LÖSNINGAR SOM TAS UPPI DENNA BRUKSANVISNING.
VARNING: ANVÄND ENDAST TILLBEHÖR
SOM REKOMMENDERAS AV TILLVERKAREN.
ANVÄNDNING AV ANDRA TILLBEHÖR KAN VARAFARLIGT, DE KAN ORSAKA ALLVARLIG SKADA PÅSYRGASKONCENTRATORN OCH DET UPPHÄVERGARANTIN.
VARNING: TILLVERKAREN REKOMMENDERAREN ALTERNATIV RESERVSYREKÄLLA I HÄNDELSE
AV STRÖMAVBROTT, LARMTILLSTÅND ELLER
MEKANISKT FEL. RÅDFRÅGA DIN LÄKAREELLER ÅTERFÖRSÄLJARE MED AVSEENDEPÅ NÖDVÄNDIGT RESERVSYSTEM.
VARNING: SE TILL ATT VENTILERNA FÖRLUFTINTAG OCH LUFTUTTAG INTE RISKERAR
ATT BLOCKERAS. TAPPA INTE OCH STICK INTE
IN NÅGRA OBJEKT I NÅGRA AV ENHETENSÖPPNINGAR. DETTA KAN LEDA TILL ATTSYRGASKONCENTRATORN ÖVERHETTAS OCH
ATT PRESTANDAN FÖRSÄMRAS.
VARNING: RÖKNING I SAMBAND MEDSYRGASANVÄNDNING ÄR DEN FRÄMSTAORSAKEN TILL BRANDSKADOR OCH DÄRTILL
RELATERADE DÖDSFALL. DU MÅSTE FÖLJADESSA SÄKERHETSVARNINGAR:VARNING: TILLÅT INTE RÖKNING, LJUSLÅGORELLER ÖPPEN ELD I SAMMA RUM SOM DENNAENHET ELLER SYREFYLLDA TILLBEHÖR.
VARNING: RÖKNING I SAMBAND MED
ANVÄNDNING AV NÄSKANYL KAN ORSAKABRÄNNSKADOR I ANSIKTET OCH EVENTUELLTDÖDSFALL.
VARNING: DESSA 3 STEG KAN RÄDDA
DITT LIV OM DU RÖKER: STÄNG AVSYRGASKONCENTRATORN, TA AV NÄSKANYLEN
OCH LÄMNA RUMMET DÄR ENHETEN STÅR.

PN MN232-C4 L | Bruksanvisning SV- 315

FreeStyle Comfort
VARNING: SKYLTAR MED ”RÖKNING EJTILLÅTEN – SYRGAS ANVÄNDS” MÅSTE
VARA VÄL SYNLIGA I HEMMET ELLER DÄRSYRGAS ANVÄNDS. ANVÄNDARE OCH DERASVÅRDGIVARE MÅSTE INFORMERAS OM FARORNAMED RÖKNING I NÄRHETEN AV ELLER UNDERANVÄNDNING AV MEDICINSK SYRGAS.
VARNING: ANVÄND INTE SYRGASKONCENTRATORNI NÄRHETEN AV LÄTTANTÄNDLIGA GASER. DET
KAN LEDA TILL SNABB FÖRBRÄNNING OCH
ORSAKA EGENDOMSSKADA, KROPPSSKADA
ELLER DÖDSFALL.
VARNING: OM MAN TAR AV NÄSKANYLEN OCH
LÄGGER DEN PÅ KLÄDER, SÄNGKLÄDER,SOFFOR ELLER ANDRA MJUKA MATERIAL KAN
BRAND PLÖTSLIGT UPPSTÅ OM NÄSKANYLENUTSÄTTS FÖR CIGARETTGLÖD, EN VÄRMEKÄLLAELLER LÅGA.
VARNING: LÄMNA INTE NÄSKANYLEN PÅ
ELLER UNDER KLÄDER, SÄNGKLÄDER ELLERSTOLSDYNOR. OM ENHETEN ÄR PÅSLAGEN MEN
INTE ANVÄNDS KAN SYRET GÖRA MATERIALET
BRANDFARLIGT. SE TILL ATT STRÖMMEN ÄR AV
NÄR SYRGASKONCENTRATORN INTE ANVÄNDS.
VARNING: GODKÄNDASYRGASTILLFÖRSELTILLBEHÖR ÄR UTFORMADEFÖR ATT MINSKA SPRIDNINGEN AV BRAND FÖRPATIENTENS SÄKERHET. ANVÄNDNING AV ICKEGODKÄNDA TILLBEHÖR KAN ÖKA RISKEN FÖRPERSON- ELLER BRANDRELATERADE SKADOR.
VARNING: SYRGASKONCENTRATORN MÅSTE
VARA AVSTÄNGD OCH BORTKOPPLAD FRÅNALL EXTERN STRÖMFÖRSÖRJNING INNANBATTERIET AVLÄGSNAS.
VARNING: SMÅDELAR KAN UTGÖRAEN POTENTIELL KVÄVNINGSRISK.
VARNING: UNDVIK KONTAKT MED
FRÅNLUFTSVENTILEN I MER ÄN 10 SEKUNDER,EFTERSOM DEN KAN VARA VARM.
VARNING: VIND ELLER KRAFTIGT LUFTDRAG
KAN HA NEGATIV INVERKAN PÅ EN KORREKTSYRGASBEHANDLING.
VARNING: DENNA PRODUKT KAN UTSÄTTA DIG
FÖR KEMIKALIER, BLAND ANNAT NICKEL, SOM I
STATEN KALIFORNIEN I USA HAR KLASSATS SOM
CANCERFRAMKALLANDE. MER INFORMATIONFINNS PÅ WWW.P65WARNINGS.CA.GOV.

316 - SV PN MN232-C4 L | Bruksanvisning

VAR FÖRSIKTIG: Koppla alltid ur nätaggregatet från väggen innan du kopplar ur
den från syrgaskoncentratorn.
VAR FÖRSIKTIG: Placera inte koncentratorn så att
det är svårt att komma åt nätsladden.
VAR FÖRSIKTIG: Placera alltid syrgasslangarna
och nätsladdarna så att det inte finns risk för att
någon snubblar eller stryps.
VAR FÖRSIKTIG: Om syrgaskoncentratorn
används med nätaggregatet för likström i en bil,båt eller andra likströmskällor, se till att fordonet
har startats och är igång innan du kopplar insyrgaskoncentratorn. Underlåtenhet att följa dessa
instruktioner kan leda till att nätaggregatet inte
förser syrgaskoncentratorn med ström.
VAR FÖRSIKTIG: När bilen som du använder
syrgaskoncentratorn i är avstängd kopplar duur och tar bort enheten från bilen. Förvara intesyrgaskoncentratorn i en mycket varm eller kallbil eller i andra liknande miljöer med hög eller låg
temperatur.
Obs! Portabel och mobil radiofrekvensutrustning för
kommunikation kan påverka medicinsk elektronisk
utrustning.
VAR FÖRSIKTIG: Lämna INTE
syrgaskoncentratorn eller nätaggregatet
inkopplat i bilen när tändningenär i AVSTÄNGT läge. Det kan tömma
fordonets batteri.
VAR FÖRSIKTIG: Om syrgaskoncentratorn används
eller förvaras utanför det normala driftstemperaturområdet och omgivningstryckområdet (altitud)kan prestandan påverkas, batteriets drifttid minska
och/eller batteriets laddningstid öka. Se avsnittet
med specifikationer i den här bruksanvisningen för
information om temperaturgränser vid förvaringoch drift.
VAR FÖRSIKTIG: För syrgaskoncentratorer som ärutrustade med batterier: Förvara på en sval och torr
plats för att säkerställa livslängden på ditt batteri.
Förvaring av syrgaskoncentratorn under längre
perioder i höga temperaturer eller med ett fulladdat
eller helt urladdat batteri kan förkorta batteriets
totala livslängd. Försök inte att öppna batteriet;
det innehåller inga delar som kan repareras av
användaren. Håll batteriet borta från barn.
VAR FÖRSIKTIG: Använd ENDAST de batterier
som tillverkaren tillhandahåller. Se till att batteriet
kasseras på ett korrekt sätt genom att kontaktaåterförsäljaren eller de lokala myndigheternaangående bestämmelser om kassering.

FreeStyle Comfort
Obs! För att minska skadeverkningarna till följd av
fel på koncentratorn eller strömförsörjningen bör
leverantören av utrustningen bedöma behovet av
reservsyrgas under installationen och med jämna
mellanrum om dessa förhållanden ändras:- Patientens tillstånd
- Patientens boendemiljö- Möjligheten att förse patienten med reservförråd
med kompletterande syrgas.
Obs! Se till att kanylen är helt instucken och ordentligt
fastsatt. Då kan syrgaskoncentratorn upptäcka
inandningar och ge syretillförsel på ett korrekt sätt.
Under inhalation kan du höra eller känna hur syre
flödar till näskanylens mynningar.

Obs! Om syrgaskoncentratorn under en längre tid har
förvarats utanför de normala driftsförhållanden som
visas i specifikationstabellen ska du låta den återfå
normal driftstemperatur innan aktivering.
Obs! Leverantören av utrustningen bör bedöma
behovet av reservsyrgas under installationen och med
jämna mellanrum om dessa förhållanden ändras:- Patientens tillstånd
- Patientens boendemiljö- Möjligheten att förse patienten med reservförråd
med kompletterande syrgas.

Obs! Litiumbatterier kan förlora sin kapacitet
permanent om de utsätts för höga temperaturer
när de är fulladdade eller helt urladdade. Vidlångtidsförvaring bör batterierna ha 25–50 % laddningoch förvaras vid temperaturer mellan -20 °C till
+20 °C (-4 °F till +68 °F).
Obs! Använd inte syrgaskoncentratorn utan att
luftintagsfiltret sitter på plats. Om ett andra filter
tillhandahålls, sätt i ersättningsfiltret innan du rengör
det smutsiga filtret. Rengör det smutsiga filtret i envarm tvålvattenlösning och torka det ordentligt innan
det används.
Obs! Tillverkaren rekommenderar inte att utrustningensteriliseras.
Obs! Det är aldrig någon risk att du tömmer ett rumpå syrgas när du använder syrgaskoncentratorn.
VARNING: SYRGASKONCENTRATORNFÅR INTE ANVÄNDAS ELLER ANSLUTAS
TILL ETT EXTERNT STRÖMUTTAG(VÄXELSTRÖM/LIKSTRÖM) UTAN ETTINSTALLERAT BATTERI. FÖRSÖK INTE
ATT SLÅ PÅ ENHETEN ELLER ANSLUTADEN TILL EXTERN STRÖM NÄR
BATTERIET ÄR BORTTAGET.
BATTERIET FÅR ENDAST TAS BORT NÄR
ENHETEN ÄR AVSTÄNGD OCH FRÅNKOPPLADFRÅN NÄTAGGREGATET. FÖRSÖK INTE ATTTA BORT KRAFTENHETEN NÄR ENHETENFORTFARANDE ÄR PÅSLAGEN ELLER ANSLUTEN
TILL NÄTAGGREGATET.

PN MN232-C4 L | Bruksanvisning SV- 317

FreeStyle Comfort
Specifikationer
FreeStyle ComfortMått (med ett batteri)25,4 x 18,5 x 7,9 cm (10,0 x 7,3 x 3,1 tum)Mått (med dubbelt batteri)27,9 x 18,5 x 7,9 cm (11,0 x 7,3 x 3,1 tum)

Vikt (med ett batteripaket)

2,3 kg (5 pund)

Vikt (med dubbelt batteripaket)

2,7 kg (6 pund)Nominell ljudnivå***39,93 dB(A) ± 0,42 dB(A) @ 2

Flödesinställningar

Pulsinställningar 1, 2, 3, 4, 5

Inställning för O2-utflöde

Inställning 1 – 210 ml/min
Inställning 2 – 420 ml/min
Inställning 3 – 630 ml/min
Inställning 4 – 840 ml/min
Inställning 5 – 1 050 ml/min([@20 andetag/min STPD]
Alla volymer +/- 15 %)Syrekoncentration*

90 % (+5,5 % / -3 %)Syrgasövervakning

Ja

Max. utloppstryck<30 PSI

Doseringskänslighet
Växelström
Likström
Typ av batteri
Batterikapacitet
Batteriladdningstid
Batteripaketets drifttid
Driftstemperatur**> -0,1 cm H2O (justerbar)
100–240 VAC, 50–60Hz, max 120 W
11–18 VDC (max 10 A)
Litiumjon
Encell: 6 700 mAh; Dubbelcell: 13 400 mAh
Ett batteri: 3,5 timmar
Dubbelt batteri: 6,0 timmar
Drifttid för ett batteri: Inställning 2 – upp till 4 timmar
Drifttid för dubbelt batteri: Inställning 2 – upp till 8 timmar
5 °C till 40 °C (41 °F till 104 °F)

Transport- och förvaringstemp. (Enhet)-25 °C till +70 °C (-13 °F till +158 °F) upp till 3 048 m (10 000 fot)Transport- och förvaringstemp. (batteri)-25 °C till +70 °C (-13 °F till +158 °F) upp till 3 048 m (10 000 fot)

Drifthöjd**-381 till 3 048 m (-1 250 till 10 000 fot) (testats för 700–1 060 hPa)

Luftfuktighet vid drift
Luftfuktighet vid förvaring
Begränsad garanti

15–95 % relativ luftfuktighet (ej kondenserande)
0–90 % icke-kondenserande
Se garantiförklaring* Baserat på ett atmosfäriskt tryck på 14,7 psi (101 kPa) vid 21 °C (70 °F)** Enheten fungerar i alla miljöer så länge miljön är vid specificerade temperatur- och fuktighetsintervall. Användningutanför dessa driftspecifikationer kan begränsa koncentratorns förmåga att uppnå den specificerade syrgaskoncentrationen
vid flödeshastigheter med ett högre literantal. Uppvärmningsskärmen kan visas ända tills koncentratorn har nåttmålkoncentrationen eller i upp till 15 minuter. Fortsätt att använda enheten under den här tiden.
*** Ljudnivå uppmätt per testmetod nr. 14-1 10/2018 MDS-Hi.
Den förväntade livslängden är minst fem år.

318 - SV PN MN232-C4 L | Bruksanvisning

FreeStyle Comfort
Bruksanvisning
1. Placera FreeStyle Comfort på en lämplig plats
för användning.
a. Koncentratorn ska vara välventilerad så att
luftintagsfiltren och luftutloppet inte täcks.
b. Koncentratorn ska placeras så att det ärlätt att se och höra alla hörbara och synliga
indikatorer och larm.
c. Kontrollera att luftintagsfiltren sitter påplats innan enheten används. Kontakta
leverantören av utrustningen om det saknas
ett filter på enheten.
2. Installera ett batteri i koncentratorn.
a. Rikta in batteriet med bottenhöljet frånFreeStyle Comforts mitt.

3. Anslut nätaggregatet för växelström eller
likström till strömkontakten som sitter påframsidan, längst ner till höger på enheten.

4. Anslut en näskanyl till syrgasutloppets
munstycke. Se till att anslutningen sitter
ordentligt före användning.
OBS! Användning av andra kanyler än de som
rekommenderats kan påverka prestandan.

b. Skjut batteriet inåt tills du hör ett klick som
bekräftar att batteriet låsts på plats.

Obs! Till återförsäljare: Följande tillbehör för
syrgashantering rekommenderas för användning med
FreeStyle Comfort:• Näskanyl med 2,1 m (7 fot) slang (maximalt LPM:
6 LPM) – CAIRE artikelnummer 5408-SEQ• Ytterligare information om rekommenderade tillbehör
finns på www.caireinc.com.
5. Slå på FreeStyle Comfort genom att trycka påströmknappen och välja den flödesinställning
som ordinerats med hjälp av knappen (+) för att
öka eller knappen (-) för att minska flödet.

PN MN232-C4 L | Bruksanvisning SV- 319

FreeStyle Comfort

Ett kort ljudlarm avges för att indikera att enheten
har slagits på.
Obs! När FreeStyle Comfort först slås på är
standardflödet inställt på 2. Om inställningen ändras
kommer enheten att startas med den föregående
flödesinställningen.
6. Placera näskanylen i ansiktet enligt de medföljandeanvisningarna från tillverkaren och andas normalt.
Du ska känna att en puff av syrgas tillförs genom
näskanylen varje gång du andas in.
7. Om du vill ändra flödesinställningen trycker
du på knappen (+) för att öka flödet eller påknappen (-) för att minska flödet till önskad
hastighet. Det är normalt att det låter annorlunda
när du byter flödesinställning.

icke-autoSAT-pulsvolym (ml)*Inställning
Andnings- 1
2
3
4
frekvens
15
14,0
28,0
42,0
56,0
20
10,5
21,0
31,5
42,0
25
8,4
16,8
25,2
33,6
30
7,0
14,0
21,0
28,0
35
6,0
12,0
18,0
24,0
40
5,3
10,5
15,8
21,0

5
70,0
52,5
42,0
35,0
30,0
26,3

8. Stäng av FreeStyle Comfort genom att trycka
på strömknappen.
Obs! Displayen försätts i viloläget och dämpas efter
20 sekunder. Tryck på valfri knapp för att aktivera
skärmen från viloläget.
9. Avancerade funktioner beskrivs i avsnittet
Anvisningar för menynavigering

PRODUKTENS SPECIALFUNKTIONER
autoSAT®
CAIRE:s egenutvecklade autoSAT-funktion gör
det möjligt för FreeStyle Comfort att tillföra en
konstant syrgasvolym (bolus) per andetag, upp tillen maximal andningsfrekvens. När användarens
andningsfrekvens ökar under aktiviteter som t.ex.
gång anpassar autoSAT-funktionen enhetens drift
för att bibehålla den inställda bolusvolymen.
UltraSense®
Den känsliga pulsdostriggern känner av
tryckförändringen under andningen och utlöser
syrgaspulsen vid inandning.

autoSAT-pulsvolym (ml)*Inställning
Andnings- 1
2
3
4
5
frekvens
15
10,5
21,0
31,5
42,0
52,5
20
10,5
21,0
31,5
42,0
52,5
25
10,5
21,0
31,5
42,0
42,0
30
10,5
21,0
31,5
35,0
35,0
35
10,5
21,0
30,0
30,0
30,0
40
10,5
21,0
26,3
26,3
26,3
*Toleransen är +/- 15 %

320 - SV PN MN232-C4 L | Bruksanvisning

autoDOSE®
Om ett larm om utebliven andning inträffar utlöser
den här funktionen syrgaspulser motsvarande
20 andetag per minut med enhetens aktuellaflödesinställning tills larmstatus har åtgärdats.

FreeStyle Comfort
Nätaggregat
FreeStyle Comfort kan drivas på tre sätt:• Laddningsbart batteri• Växelström• Likström
Information om hur dessa komponenter fungerar
ges i de följande avsnitten.
VARNING: ANVÄND ENDAST DENÄTAGGREGAT SOM BESKRIVS I DENHÄR ANVÄNDARHANDBOKEN. OM
NÄTAGGREGAT ELLER BATTERIPAKETFRÅN EN ANNAN TILLVERKARE ÄN
CAIRE ANVÄNDS KAN ENHETENSKADAS OCH FUNGERA FELAKTIGT.

TA UT BATTERIET

Tryck ner batterifrigöringsknappen bort frånFreeStyle Comfort. HÅLL knappen intryckt och dra
ut batteriet ur koncentratorn.
VARNING: SYRGASKONCENTRATORNMÅSTE VARA AVSTÄNGD OCH
FRÅNKOPPLAD FRÅN EXTERN STRÖM
INNAN BATTERIPAKETET TAS UT.

BATTERIPAKETFreeStyle Comfort kan drivas med det
laddningsbara batteripaketet som medföljer
utrustningen. Koncentratorn har inga driftsmässiga
begränsningar när den drivs med batteripaketet.
Alla flödeshastigheter är tillgängliga vid batteridrift.

KONTROLLERA BATTERIPAKETETSLADDNING
INSTALLERA BATTERIET

Rikta in batteriet med bottenhöljet från FreeStyle
Comforts mitt. Skjut batteriet inåt tills du hör ett
klick som bekräftar att batteriet låsts fast på plats.

Se batterisymbolen och uppskattad återstående
batteritid på displayen för att kontrollera
batteriladdningen när det är installerat
i FreeStyle Comfort.
80–100 %60–80 %

40–60 %

20–40 %20 %PN MN232-C4 L | Bruksanvisning SV- 321

FreeStyle Comfort
Obs! Om strömkällan är ansluten visas en uttagsindikator
i stället för kvarvarande laddningsprocent.
Om du vill kontrollera batteriladdningen när
batteriet inte är installerat i FreeStyle Comfort
placerar du batteriet plant och trycker på TESTknappen. Indikatorlamporna för batterimätaren(25–100 %) tänds till höger om TEST-knappen
för att visa batteripaketets laddningsnivå.

DRIFTTID VID BATTERIDRIFT

När extern växelström eller likström kopplas från,växlar koncentratorn automatiskt över till batteridriftom ett laddat batteri är installerat. I tabellen nedan
visas normal drifttid för ett nytt batteripaket.
Batteripaketets drifttid förblir konstant oavsett din
andningsfrekvens.
Pulsinställning

Drifttid för ett
batteri (8 celler)

Drifttid för dubbelt
batteri (16 celler)

1

upp till 8 timmar

upp till 16 timmar

2

upp till 4 timmar

upp till 8 timmar

3

upp till 3 timmar

upp till 6 timmar

4

upp till 2,25 timmar

upp till 4,5 timmar

5

upp till 2 timmar

upp till 4 timmar

NORMAL BATTERILADDNINGSTIDFÖRSTA LADDNINGEN

Det nya batteriet som medföljer din FreeStyle
Comfort är bara delvis laddat när det levererasfrån fabriken. Innan du använder enheten för
första gången ska du ladda batteripaketet helt med
växelström eller likström.

ALLMÄN INFORMATION OM LADDNINGFreeStyle Comfort laddar batteripaketet när det
är installerat i koncentratorn och koncentratorn
är ansluten till en extern växelströms- eller
likströmskälla (förutom på ett flygplan).
Batteriet är fulladdat när det visar 100 %.
Alternativt kan batteripaketet laddas med
bordsladdaren (tillval).

322 - SV PN MN232-C4 L | Bruksanvisning

Det tar vanligtvis 3,5 timmar för ett batteri och
6,0 timmar för ett dubbelt batteri för att helt ladda
batteripaketet när det är helt urladdat.
Obs! Batteriet kanske inte börjar laddningen direktom batteripaketet är helt urladdat.

BATTERIETS LIVSLÄNGD

FreeStyle Comfort-batterier är utformade för att
hålla för 300 laddnings-/urladdningscykler.

FreeStyle Comfort
Nätaggregat för växelström och
likström
FreeStyle Comfort levereras som standard med
nätaggregat för växelström och likström.
I nätaggregatet för växelström ingår nedanstående:	a. Växelströmdel som drivs på 100–240 V,
50–60 Hz.
	b. Växelströmkabel för anslutning till
växelströmdel och lämpligt växelströmsuttag.

Nätaggregatet för likström består av en sladd som
ansluts direkt till enheten och likströmssuttaget.

2. Anslut nätsladden till växelströmdelen.

VÄXELSTRÖM

Växelström kan användas hemma och överallt där
det finns vanliga nätströmsuttag. Freestyle Comfort
har full funktionalitet när den drivs med växelström.
FreeStyle Comfort laddar även batteripaketet när
som helst där växelström finns tillgängligt (utom i
flygplan). Batteripaketet laddas oavsett om FreeStyle
Comfort är påslagen eller avstängd.
VARNING: ANSLUT INTESYRGASKONCENTRATORNTILL EXTERN STRÖM NÄR ETTBATTERIPAKET INTE ÄR INSTALLERAT.

3. Anslut den andra änden av nätsladden till ett
vägguttag eller ett annat lämpligt eluttag.

Så här ansluter du Freestyle Comfort till växelström:
1. Anslut växelströmdelen till strömingången
på framsidan, längst ner på FreeStyle Comfort.
Tryck inåt så att anslutningen sitter säkert.

PN MN232-C4 L | Bruksanvisning SV- 323

FreeStyle Comfort
4. Slå på FreeStyle Comfort genom att trycka påströmknappen och välja den flödesinställning
som ordinerats med hjälp av knappen (+) för att
öka eller knappen (-) för att minska flödet.

4. Slå på FreeStyle Comfort genom att trycka påströmknappen och välja den flödesinställning
som ordinerats med hjälp av knappen (+) för att
öka eller knappen (-) för att minska flödet.
Obs! Om batteripaketet behöver laddas visas batterisymbolen på displayen med ett enda orangefärgat streck.

LIKSTRÖM

Likström används vid anslutning till tillbehörsuttag
i bilar, båtar och andra motorfordon. FreeStyle
Comfort har full funktionalitet när den drivs med
likström. Freestyle Comfort laddar också vid
behov batteripaketet när likström är tillgängligt.
Batteripaketet laddas både när enheten är avstängd,påslagen och när den används.
VARNING: ANSLUT INTE EXTERN STRÖMUTAN ETT BATTERIPAKET INSTALLERATI SYRGASKONCENTRATORN.
1. Starta motorfordonet.
2. Anslut nätaggregatet för likström till
strömuttaget på framsidan, längst ner
på FreeStyle Comfort. Tryck inåt så att
anslutningen sitter säkert.

3. Anslut den andra änden av likströmssladden till
ett uttag i en bil eller ett annat likströmsuttag.

324 - SV PN MN232-C4 L | Bruksanvisning

VAR FÖRSIKTIG: För att förhindra
brand eller överhettning, se till 12 V
likströmuttaget är fritt från skräp och att
adapterkontakten för nätaggregatet för
likström är ordentligt isatt och sitter fastordentligt.
VAR FÖRSIKTIG: Metallytor på nätaggregat för
likström kan bli MYCKET VARMA vid användningpå grund av den höga strömstyrkan. Låt ytornasvalna innan du rör vid dem.
VAR FÖRSIKTIG: Använd inte nätaggregatet
för likström tillsammans med förlängningskablar,
grenuttag eller adaptrar. Användning av icke
godkända tillbehör kan leda till att de överhettas
eller skadas.
VAR FÖRSIKTIG: Om enheten ansluts till ett 12 V
likströmsuttag kan fordonets batteri tömmas om
motorn inte körs, även om apparaten är avstängd.
VAR FÖRSIKTIG: Starta inte ett fordon med
starthjälp när nätaggregatet för likström är anslutet.

FreeStyle Comfort
Rengöring, skötsel och
rutinunderhåll
RUTINMÄSSIGT UNDERHÅLL

Förebyggande underhåll
Enheten är underhållsfri. Filter inuti enheten är
avsedda att hålla under enhetens hela livslängd.
Om enheten inte fungerar korrekt ska enheten
returneras till leverantören för reparation.
Service av interna komponenter innanför
höljet på FreeStyle Comfort ska utföras av
syrgasleverantören.
Användaren av enheten ansvarar för nedanstående
underhållsprocedurer.
• Varje vecka – rengöring av luftintagsfiltren
VARNING: FÖRSÖK INTE ATT ÖPPNAELLER TA BORT PLASTHÖLJET PÅSYRGASKONCENTRATORN. ENHETEN
FÅR ENDAST ÖPPNAS AV EN BEHÖRIGSERVICETEKNIKER.

LUFTINTAGSFILTER

Luft kommer in i FreeStyle Comfort genom
luftintagsfilter som sitter på framsidan, längst ner påsyrgaskoncentratorn. Filtren förhindrar att damm ochandra större partiklar i luften kommer in i enheten.
Luftintagsfiltren ska rengöras minst en gång iveckan. Syrgasleverantören kan råda dig att rengöra
dem oftare, beroende på driftsförhållandena. Följ
dessa anvisningar för att rengöra luftintagsfiltren
ordentligt.
1. Ta bort de två filtren från framsidan av FreeStyle
Comfort genom att trycka upp motsvarande flikoch dra filtren utåt.

VARNING: ANVÄND INTE SYRGASKONCENTRATORN UTAN ATT LUFTINTAGSFILTRET SITTER PÅ PLATS.

SKÖTSEL AV BATTERIPAKETET

Låt inte batteripaketet komma i kontakt med vätska.
Försök inte att demontera eller öppna batteripaketet.
Låt inte batteriet komma i kontakt med öppen eld.
Du får inte tappa batteripaketet.
Håll batteripaketet utom räckhåll för barn.
Om batteripaketet verkar vara skadat ska dukontakta syrgasleverantören innan du använder det.
Placera inte batteriet i eller i närheten av en
mikrovågsugn eller annan köksutrustning.

FÖRVARING AV BATTERIPAKETET

Syrgaskoncentratorns batteri behöver inte laddas ur
helt innan det laddas på nytt. Syrgaskoncentratorns
batteri bör laddas efter varje användning.
Lämna inte batteriet på en plats med mycket hög
värme, t.ex. i direkt solljus eller i en bil. Om du
gör det kan batteriets prestanda och livslängd
försämras. Det kan även leda till att batteriet
alstrar värme, börjar ryka, tar eld eller exploderar.
Batteripaketet ska förvaras i en stabil miljö med låg
luftfuktighet (lägre än 70 % relativ luftfuktighet)
och utan frätande gaser. För att förhindra rostska förhållanden som kan ge upphov till kondens
undvikas, exempelvis snabba växlingar i den
omgivande temperaturen.
Vid förvaring i en månad eller längre tid
rekommenderas en laddningsnivå på mellan
50 % och 75 % (två till tre indikatorlamporlyser). När batteriet laddats ska det tas ut ursyrgaskoncentratorn.

2. Tvätta båda filtren med tvål och varmt vatten.
3. Skölj filtren noggrant och avlägsna vattnet meden mjuk luddfri trasa. Se till att filtren är helt
torra innan du sätter tillbaka dem.
4. Sätt tillbaka de torra, rena filtren.
PN MN232-C4 L | Bruksanvisning SV- 325

FreeStyle Comfort
RENGÖRING

Stäng AV enheten och koppla bort den frånlikströmmen eller växelströmmen innan rengöring.
Spreja INGET direkt på ytterhöljet. Använd
en fuktig (inte blöt) trasa eller svamp. Rengör
höljet och nätaggregaten genom att fukta trasaneller svampen med ett milt rengöringsmedel
för hushållsbruk. Gör som tillverkaren av
rengöringsmedlet föreskriver.
VARNING: STÄNG AV ENHETEN OCH
KOPPLA BORT NÄTSLADDEN FRÅNELUTTAGET INNAN DU RENGÖRENHETEN FÖR ATT UNDVIKA ELSTÖTAR
OCH BRÄNNSKADOR. ENDAST DINÅTERFÖRSÄLJARE ELLER EN BEHÖRIGSERVICETEKNIKER FÅR TA BORTHÖLJET ELLER REPARERA ENHETEN.
VARNING: SÄKERSTÄLL ATT SYRGASKONCENTRATORN OCH BATTERIPAKETET INTE BLIRBLÖTA OCH ATT VÄTSKOR INTE KOMMER IN IENHETEN. DET KAN LEDA TILL FEL ELLER ATTENHETEN STÄNGS AV OCH KAN MEDFÖRA RISKFÖR ELSTÖTAR ELLER BRÄNNSKADOR.
VARNING: ANVÄND INTE VÄTSKA DIREKTPÅ ENHETEN. OLÄMPLIGA KEMISKA MEDELOMFATTAR, MEN ÄR INTE BEGRÄNSADE TILLFÖLJANDE: ALKOHOL OCH ALKOHOLBASERADEPRODUKTER, PRODUKTER BASERADE PÅ
KONCENTRERAD KLORIN (ETYLENKLORID)OCH OLJEBASERADE PRODUKTER. DESSA
PRODUKTER FÅR INTE ANVÄNDAS FÖR ATTRENGÖRA KONCENTRATORNS PLASTHÖLJEEFTERSOM DE KAN SKADA ENHETEN.
VARNING: RENGÖR KÅPAN, KONTROLLPANELEN OCH NÄTSLADDEN ENBART MED ETT MILTHUSHÅLLSRENGÖRINGSMEDEL OCH EN FUKTAD(INTE BLÖT) DUK ELLER SVAMP OCH TORKASEDAN ALLA YTOR TORRA. SE TILL ATT VÄTSKA
INTE KOMMER IN I KONCENTRATORN. VAR SÄRSKILT NOGGRANN MED SYRGASUTLOPPET DÄR
KANYLEN ANSLUTS OCH SE TILL ATT DET ÄR
FRITT FRÅN DAMM, VATTEN OCH PARTIKLAR.

326 - SV PN MN232-C4 L | Bruksanvisning

VARNING: ANVÄND EN FUKTAD (INTE BLÖT)DUK ELLER SVAMP FÖR ATT RENGÖRABATTERIPAKETET. SPREJA FÖRST ETTMILT HUSHÅLLSRENGÖRINGSMEDEL PÅ
DUKEN ELLER SVAMPEN OCH RENGÖRBATTERIPAKETETS HÖLJE.
VARNING: FÅR INTE RENGÖRAS MED ETTAUTOMATISERAT RENGÖRINGSSYSTEM.

BYTE AV KANYL

Följ alltid kanyltillverkarens anvisningar för korrekt
användning. Byt ut näskanylen eller syrgasslangenvar 3:e till 6:e månad (vid daglig användning, var
3:e månad) enligt tillverkarens rekommendation
enligt kanyltillverkarens eller syrgasleverantörens
rekommendationer. Läkaren eller syrgasleverantören
ger dig instruktioner för rengöring och byte.
Extra tillbehör för byte är tillgängliga via
syrgasleverantören.

FreeStyle Comfort
Larmtillstånd
Prioritet nr

Beskrivning
av larm

1

Systemfel
Hög temperatur påbatteriet

Display

Ljudsignal

Mer information/föreslagen åtgärd

1 pipsignal
var 20:esek

Enheten avger ett larm när batteritemperaturen börjar närma sig säkerhetsavstängningsgränsen.
Om avstängningsgränsen nås, stängs
enheten av. Om extern strömförsörjningansluts visas larmskärmen och ett ljudlarm
fortsätter i 2 min.
Testa med ett batteri som har rumstemperatur. Kontakta leverantören om problemetkvarstår.

2

Systemfel
Högtemperatur

1 pipsignal
var 20:esek

Enheten avger ett larm när den interna
temperaturen börjar närma sig avstängningsgränsen.
Om avstängningsgränsen nås stängs
enheten av. Larmskärmen visas och ett
ljudlarm fortsätter i 2 min.
Se till att det inte finns några blockeringar
i enhetens ventilationsöppningar. Kontakta
leverantören om problemet kvarstår.

3

Systemfel
Högt tryck(fel på högt
systemtryck
eller kompressorfel)

1 pipsignal
var 20:esek

4Systemfel
Flödeshastighet (OCDventilfel,
obstruktion
av gasväg)

1 pipsignal
var 30:esek

Kontakta leverantören om problemetkvarstår.

5Låg
batterinivå
Avstängning

1 pipsignal
var 30:esek

Larmet avges tills batteriet laddas ur och
enheten stängs av.

Om avstängningsgränsen nås stängs
enheten av. Larmskärmen visas och ett
ljudlarm fortsätter i 2 min.
Kontakta leverantören om problemet kvarstår.

Ladda batteriet.

PN MN232-C4 L | Bruksanvisning SV- 327

FreeStyle Comfort
Prioritet nr

Beskrivning
av larm

6

Inget
andetagdetekterat

Display

Ljudsignal

Mer information

1 pipsignal
var 10:emin

Inget andetag detekterat under 15 sekunder.
Aktiveras efter första andetaget eller när
uppvärmningsskärmen stängs av, beroendepå vad som inträffar först.
Kontrollera om kanylen har veck eller läckor.
Öka känslighetsinställningen. Kontakta
leverantören om problemet kvarstår.

7

Uppvärmningingen

Enheten börjar automatisk dosering tills ett
andetag detekteras eller uppvärmningen avslutas, beroende på vad som inträffar först.

8

Låg koncentration

1 pipsignal
var 10:emin, i upp
till 30 min

Kontakta leverantören om problemetkvarstår.

9Låg batterinivå Varning

1 pipsignal,
upprepas ej

Ladda batteriet.

10

Andningsfrekvensöverskriden

1 pipsignal,
upprepas ej

Andningsfrekvensen överskrider
40 andetag/min.

11

Systemfel
Fläktfelingen

Kontakta leverantören om problemetkvarstår.

12

SystemfelLågt tryck

ingen

Kontakta leverantören om problemetkvarstår.

13

Systemfel
Låg temp

ingenLåt enheten värmas upp till rumstemperatur. Kontakta leverantören om problemetkvarstår.

328 - SV PN MN232-C4 L | Bruksanvisning

Om dosen utlöses felaktigt ska känslighetsinställningen minskas. Kontakta
leverantören om problemet kvarstår.

FreeStyle Comfort
Prioritet nr

Beskrivning
av larm

Info

Start

Display

Ljudsignal

Mer information

1 informationspip

Skärmen visas vid påslagning och dubör höra en pipsignal som bekräftar att
signalen fungerar.
Startskärmen visar aktuell revision för fast
programvara längst ner i det högra hörnet.

Info

Skärm vid
normal drift
i autoSAT-läget (ingen
likström
ansluten)

ingen

Info

Skärm vid
normal drift
med autoSAT
inaktiverat(ingenlikström
ansluten)

ingen

Info

Ändradinställning

1 informationspip

Info

Inkoppling av
extern ström
i avstängt
läge

ingen

Info

Inkoppling av
extern ström
i driftläget

ingen

Info

Frånkoppling
av extern
ström iavstängt
läge

1 informationspip

Visar CAIRE-skärmen i 2 sekunder, följt
av batteri- och kontaktstatus i 5 sekunderinnan skärmen stängs av.

1 Visar CAIRE-skärmen i 2 sekunder och
stänger sedan av displayen.

PN MN232-C4 L | Bruksanvisning SV- 329

FreeStyle Comfort
Prioritet nr

Beskrivning
av larm

DisplayInfoFrånkoppling
av extern
ström i driftläget (medbatteri)

1 informationspip

Info

När
upp- eller
nedknappen
trycks ner
i viloläget
visas
batteri- ochladdningsstatus på
displayen

ingen

Info

Flygplansläge när enheten drivs på
batteri

1 långljudsignal

Info

Flygplansläge när
enheten är
ansluten tillström

1 långljudsignal

330 - SV PN MN232-C4 L | Bruksanvisning

Ljudsignal

Mer information

Skärmen visas i 5 sekunder och stängs
sedan av.

FreeStyle Comfort
Anvisningar för menynavigering
Menyn kan öppnas i viloläge eller driftläge. Om ett
larmtillstånd inträffar i driftläge avslutas menyn
automatiskt. Om inga knappar trycks in under
20 sekunder avslutas menyn automatiskt. I viloläge
kan du behöva trycka på menyknappen två gånger
för att aktivera enheten och sedan gå in i menyn.
Det finns fyra menyskärmar:
1. Statusskärm: Visar revisionsnummer för
firmware, drifttid och batteriurladdningscykler

Flygplansläge

2. Flygplansläge (PÅ/AV)
3. Känslighet: justerbar (1–5)
4. autoSAT (PÅ/AV)

Känslighet

Exempel på statusskärm

ANVÄNDA INSTÄLLNINGSMENYN
Du når inställningsmenyn genom att trycka på
meny-/valknappen
Tryck påi driftläge eller viloläge.

autoSATför att bläddra genom menyerna.

Statussidan

PN MN232-C4 L | Bruksanvisning SV- 331

FreeStyle Comfort
STATUSSIDAN
Statussidan för FreeStyle Comfort visar
revisionsnummer för firmware, drifttid och antal
battericykler på det anslutna batteriet.

Tryck påmedan Cancel-symbolen (Avbryt)visas för att avbryta valet av flygplansläge på.

Flygplansläge på bekräftat

Statussidan

Drifttid visas i hela timmar.
Batteriurladdningar är antalet cykler som det
anslutna batteriet har tömts och laddats.

FLYGPLANSLÄGE
FreeStyle Comfort levereras med flygplansläget
avstängt.

Upprepa ovanstående steg och växla till OFF (AV)för att stänga av flygplansläget.
Om du trycker på
när skärmen visar
Next-symbolen (Nästa) flyttar menysidan till nästa
skärmalternativ.
Flygplansläget är avsett för användning i driftläge.
Flygplansläget inaktiveras automatiskt om enheten
lämnas i viloläge med batteridrift.
Obs! Se avsnittet Aktivera flygplansläget för mer
information om flygplansläget.

KÄNSLIGHETSLÄGE

Huvudskärm i flygplansläget

Tryck påmedan Change-symbolen (Ändra)
visas för att aktivera läget. Värdet växlar då till
ordet ON (PÅ) i blått.

FreeStyle Comfort levereras från fabriken med
standardinställningen 2 för känslighetsläget och bör
inte behöva justeras under normala förhållanden.
Känslighetsinställningen kan justeras mellanvärdena 1 och 5, där 1 innebär minst känslighet och
5 mest känslighet.

Skärmen i känslighetsläge vid inställning 2
Flygplansläge på har valts

Tryck påmedan Save-symbolen (Spara) visas
för att bekräfta flygplansläge på. Värdet växlar dåtill ordet ON (PÅ) i svart.

332 - SV PN MN232-C4 L | Bruksanvisning

FreeStyle Comfort
Tryck påmedan Change-symbolen (Ändra)
visas för att justera känslighetsläget. Värdet växlar
då till nästa siffra i blått.

Inställning 3 för känslighetsläget vald

Tryck påmedan Save-symbolen (Spara) visas
för att bekräfta inställning 3 för känslighetsläget.
Värdet växlar då till siffran 3 i svart.

AUTOSAT-LÄGEFreeStyle Comfort levereras med autoSAT-läget
aktiverat.

Huvudskärm i autoSAT-läget

Tryck påmedan Change-symbolen (Ändra)
visas för att inaktivera läget. Värdet växlar då till
ordet OFF (AV) i blått.

medan Cancel-symbolen (Avbryt) ärTryck påsynlig för att avbryta ändringen av känslighetsläge.

autoSAT-läge AV har valts

Inställning 3 för känslighetsläget bekräftad

Tryck påmedan Save-symbolen (Spara) visas
för att bekräfta autoSAT-läge på. Värdet växlar dåtill ordet OFF (AV) i svart.
medan Cancel-symbolen (Avbryt)
Tryck påvisas för att avbryta valet av autoSAT-läge på.

Inställning 5 (max) för känslighetsläget bekräftad

Om du trycker på
när skärmen visar
Next-symbolen (Nästa) flyttar menysidan till nästa
skärmalternativ.

autoSAT-läge av bekräftat

Upprepa ovanstående steg och växla till ON (PÅ)
för att aktivera autoSAT-läget.
Om du trycker på
när skärmen visar
Next-symbolen (Nästa) flyttar menysidan till nästa
skärmalternativ.

PN MN232-C4 L | Bruksanvisning SV- 333

FreeStyle Comfort
Felsökning
Problem

Trolig anledning

Lösning

FreeStyle Comfort visar
ett larm och avger enåterkommande ljudsignal.

Se tabellen överlarmförhållanden.

Se tabellen över larmförhållanden.

Syrgaskoncentratorn startar
inte.

Batteriet är urladdat.

Förse enheten med ström genom växel- eller
likströmsuttag.

Inte ansluten till en externströmkälla.
Allmänt fel.

Se till att de externa kontakterna sitter säkert.
Utför en fulleffektscykel genom att ta ut batterietoch koppla från eventuell extern växelströmseller likströmskälla. Sätt sedan tillbaka batteriet
och effektcykeln är klar. Anslut vid behov den
externa strömkällan igen.
Kontakta leverantören av utrustningen och byt
vid behov till en annan syrgaskälla.

Alla övriga problem.

Utför en fulleffektscykel genom att ta ut batterietoch koppla från eventuell extern växelströmseller likströmskälla. Sätt sedan tillbaka batteriet
och effektcykeln är klar. Anslut vid behov den
externa strömkällan igen.
Byt till en annan tillgänglig syrgaskälla och
kontakta leverantören av utrustningen.

Obs! Vissa larm gör så att enheten stängs av.
Obs! Under en total strömförlust återställs larmsystemet (eventuella tidigare larm rensas) och ett larm visas endastom tillståndet kvarstår när strömmen återställs.

334 - SV PN MN232-C4 L | Bruksanvisning

FreeStyle Comfort
Information inför resor
FreeStyle Comfort följer FAA:s riktlinjer för
användning ombord på trafikflygplan. Före varje
resa (med flyg, tåg, kryssningsfartyg osv.)
ska du vidta följande åtgärder:• Kontrollera att koncentratorn är i ett
välfungerande skick.
• Kontakta researrangören minst 48 timmar föreavresa och informera dem om att du kommer att
resa med FreeStyle Comfort.
Du bör alltid ha med extra batterier när du reser.
Vid flygresor ska du se till att drifttiden med
batteri räcker för tiden på marken (incheckning,säkerhetskontroll osv.) och mellanlandningar.
De flesta flygbolag kräver att du har ett tillräckligt
antal fulladdade batteripaket för att driva enheten
under minst 150 % av den förväntade maximala
restiden (flygtider, tid på marken före ochefter flygresan, anslutningar samt oförutsedda
förseningar).
Obs! Vid flygresor bör extra batterier förvarasi en säker transportväska.

FAA-BESTÄMMELSER

Reservbatterier måste skyddas var för sig för
att förhindra kortslutning. De kan placerasi originalförpackningen eller isoleras på annat sätt,t.ex. genom att exponerade terminaler tejpas för
eller varje batteri placeras i en separat plast- eller
skyddspåse, och bör endast transporteras som
handbagage.

1. Slå på POC med ENDAST batteridrift,utan någon extern strömkälla ansluten.

2. Aktivera flygplansläget genom någon av
metoderna nedan.
a. Följ anvisningarna för aktivering av
flygplansläge i avsnittet Anvisningar för
menynavigering.

b. Håll in både upp- och nedknappen
i 5 sekunder tills du hör en lång pipsignal
och ser att flygplanssymbolen visas.

AKTIVERA FLYGPLANSLÄGET
Batteridrift
Den bärbara syrgaskoncentratorn kan användas
på ett flygplan ENDAST med batteridrift undernormala driftförhållanden, när en extern strömkälla
inte är ansluten
Extern ström
På ett flygplan, om ett eluttag är tillgängligt, kan
användas och dess läge är lämpligt för användningsom en POC-strömkälla, kan det fungera som en
backup till POC-batterier. I det fallet måste dufölja nedanstående rekommenderade förfarande
för övergång från batteri till flygplanets elsystem:

3. Koppla in den bärbara syrgaskoncentratorn när
flygplanssymbolen visas.

Obs! På grund av flygplanets begränsningar laddas
inte batterierna i flygplansläget.

PN MN232-C4 L | Bruksanvisning SV- 335

FreeStyle Comfort
TillbehörSÄTTA I ENHETEN I TRANSPORTVÄSKAN

För korrekt funktion och säkerhet ska du endast
använda nedanstående tillbehör från CAIRE somlevereras via din syrgasleverantör. Om du använder
andra tillbehör än de som räknas upp nedan
kan koncentratorns prestanda och/eller säkerhetpåverkas.

1. Skjut in enheten till botten av väskan och rikta infilter- och batteriöppningarna korrekt.

Standardtillbehör till FreeStyle Comfort
Transportväska

MI396-1

Ryggsäck

MI459-1

Axelrem

MI397-1

Nätaggregat för växelström (allasladdar medföljer)

PW036-1S

Likströmssladd

CD041-1

Externt partikelfilter, grovt (höger)FI226-1

Externt partikelfilter, grovt (vänster)FI227-1

2. Stäng blixtlåset runt botten av väskan.

Extra tillbehör som finns tillgängliga
Extra 8-cellsbatteri

BT034-1S

16-cells batteripaket

BT035-1S

Bordsladdare för batteripaket

BT036-1

Tillbehörsväska

MI372-2

Obs! Fler alternativ för landsspecifika nätsladdar
kan vara tillgängliga där detta anges ovan. Kontakta
CAIRE eller din syrgasleverantör om du behöver
beställa andra alternativ.

Anvisningar för transportväskan
Sätt i och förankra FreeStyle Comfort-enheten
i transportväskan enligt anvisningarna nedan.

336 - SV PN MN232-C4 L | BruksanvisningFÄSTA AXELREMMEN
1. Axelremmen kan fästas antingen direktpå enheten eller på transportväskan.
2. Snäpp fast remmen i ringen på enheten eller
väskan.

FreeStyle Comfort
Kassering och återanvändningÅterlämna alltid FreeStyle Comfort inklusive alla
komponenter till leverantören av utrustningen
för korrekt avfallshantering. Du kan även kontakta
ditt lokala förvaltnings- eller kommunkontor
för instruktioner om korrekt avfallshantering
av batteriet. Enheten kan användas igen av olika
personer men måste återlämnas till leverantören
av utrustningen för rekonditionering.

WEEE och RoHS
Syftet med denna symbol är att påminnaägaren av utrustningen att lämna tillbaka den till enåtervinningsanläggning när den ska kasseras, enligt
WEEE-direktivet om kassering av elektriskt avfall(Waste Electrical and Electronic Equipment).

Material som har direktkontakt med
användaren• Koncentratorns hölje: PC/ABS• Koncentratorns kontrollpanel: Polyester EBG7
eller likvärdigt• Syrgasutlopp, munstycke: Aluminium• Enhetsmärkning: Lexan• Kabelanslutningar: Polykarbonat/vinylklorid• Nätsladd: PVC, metall• Strömförsörjning: Lexan 940 (polykarbonat)• Batteripaket: PC/ABS• Batteripaket, Strömförsörjningsmärkning:
Polyesterfilm• Koncentratorns transportväska: Cordura, neopren,nylonfoder• Rem: Polypropylen, Acetyl, Vinyl, Cordura,
neoprenVåra produkter uppfyller kraven i RoHS-direktivet(begränsning av användningen av vissa farligaämnen i elektriska och elektroniska produkter).
De innehåller inte mer än spårelement av bly ochandra farliga material.
VAR FÖRSIKTIG: Kontakta leverantöreneller de lokala myndigheterna för
anvisningar om korrekt avfallshantering
av batterier.
VAR FÖRSIKTIG: Kassera alltid molekylsikt
i enlighet med alla federala, statliga och lokala
miljöbestämmelser.

PN MN232-C4 L | Bruksanvisning SV- 337

FreeStyle Comfort
EMC-tester
Vägledning och tillverkarens deklaration – elektromagnetiska emissioner
FreeStyle Comfort är avsedd för användning i den elektromagnetiska miljö som specificeras nedan.
Kunden eller användaren av FreeStyle Comfort ska säkerställa att den används i en sådan miljö.
Emissionstester
RF-emissioner
CISPR 11
RF-emissioner
CISPR 11
Harmoniska emissioner
IEC 61000-3-2
Spänningsvariationer/
flimmer IEC 61000-3-3Överensstämmelse

Elektromagnetisk miljö – vägledning

Grupp 1FreeStyle Comfort använder endast radiofrekvensenergi förinterna funktioner. Därför är RF-emissionerna väldigt låga och det
är inte sannolikt att de inverkar störande på elektronisk utrustningi närheten.

Klass B
Klass AÖverensstämmer

338 - SV PN MN232-C4 L | Bruksanvisning

FreeStyle Comfort lämpar sig för användning i alla inrättningar,
inklusive hemmiljöer och sådana som är direktanslutna till det
allmänna elnätet med lågspänning som försörjer byggnader som
används som bostäder.

FreeStyle Comfort
Vägledning och tillverkarens deklaration – elektromagnetisk immunitet

FreeStyle Comfort är avsedd för användning i den elektromagnetiska miljö som specificeras nedan. Kunden eller
användaren av FreeStyle Comfort ska säkerställa att den används i en sådan miljö.
Immunitetstest
IEC 60601-testnivå
Överensstämmelsenivå Elektromagnetisk miljö – vägledning
Golven ska vara av trä, betong eller
Elektromagnetisk±8 kV kontakt±8 kV kontaktkakel. Om golvet är täckt med ett
miljö – vägledning
syntetiskt material bör den relativa±15 kV lufturladdning±15 kV lufturladdning
IEC 61000-4-2
luftfuktigheten vara minst 30 %.
±2 kV för strömförsörj±2 kV för strömförsörjSnabba elektriska
ningsledningar
Nätströmmen ska vara samma somningsledningar
transienter/i en typisk professionell sjukvårdsinrätttransientskurar
ning eller hemsjukvårdsmiljö.
±1
kV
för
in-/
IEC 61000-4-4
Ej tillämpligtutgångsledningar±1 kV ledning(ar) till
ledning(ar)Toppar
IEC 61000-4-5±2 kV normalläge för
växelströmledningar±1 kV differentialläge för
växelströmledningar±1 kV ledning(ar)till ledning(ar)Nätströmmen ska vara samma somi en typisk professionell sjukvårdsinrättning eller hemsjukvårdsmiljö.

±2 kV normalläge för
I/O-kopplingar utomhus
0 % UT i 0,5 cykel
0 % UT i 0,5 cykel(0°, 45°, 90°, 135°,(0°, 45°, 90°, 135°, 180°,
Nätströmmen ska vara samma
180°, 225°, 270°, 315°) 225°, 270°, 315°)
som i en typisk professionell
Spänningsfall, korta
sjukvårdsinrättning eller
avbrott och spän0 % UT i 1 cykel (0°)
0 % UT i 1 cykel (0°)
hemsjukvårdsmiljö. Om användarenningsvariationer hos
av FreeStyle Comfort har behovströmförsörjningens
av kontinuerlig drift även under
70
%
U
70
%
U
(30
%
fall
i
U
)
(30
%
fall
i
U
)
inloppsledningar
TT
T
T
strömavbrott, rekommenderas att
i 25/30 cykler (0°)
i 25/30 cykler (0°)
IEC 61000-4-11
FreeStyle Comfort drivs via avbrottsfri
strömförsörjning eller batteri.
0 % UT i 250/300 cykler 0 % UT i 250/300 cykler
(0°)
(0°)
Magnetfält som uppstår på grund
Spänningsfrekvenav spänningsfrekvens ska ligga
sens (50/60 Hz)
3 A/m
3 A/m
på nivåer som är typiska för enmagnetfält,
professionell sjukvårdsinrättning och
IEC 61000-4-8
hemsjukvårdsmiljö.
OBS! UT är växelströmsspänningen innan testnivån tillämpas.
OBS! I vissa fall kan Freestyle Comfort återställas efter en betydande elektrostatisk urladdning och kan kräva
en fulleffektscykel för att återuppta normal drift. Se felsökningsavsnittet för mer information.

PN MN232-C4 L | Bruksanvisning SV- 339

FreeStyle Comfort
Vägledning och tillverkarens deklaration – elektromagnetisk immunitet

FreeStyle Comfort är avsedd för användning i den elektromagnetiska miljö som specificeras nedan. Kunden eller
användaren av FreeStyle Comfort ska säkerställa att den används i en sådan miljö.
Immunitetstest IEC 60601-testnivå
Överensstämmelsenivå Elektromagnetisk miljö – vägledning
10 Vrms
Portabel och mobil RF-kommunikationsLedningsburen RF 10 Vrms
utrustning ska inte användas närmare
150 kHz till 80 MHz
IEC 61000-4-6någon del av FreeStyle Comfort, inklusive
kablarna, än det rekommenderade minsta
separationsavståndet som beräknats
med den ekvation som är tillämplig för
10 V/m
10 V/mUtstrålad RF
sändarens frekvens.
80 MHz till 2700 MHz
IEC 61000-4-3
Rekommenderat separationsavstånd

d = 1.2 P
d = 1.2 P = 80 MHz till 800 MHz
d = 1.2 P = 800 MHz till 2 700 MHz
där P är sändarens maximala uteffekt
i watt (W) enligt sändartillverkaren
och d är det rekommenderade
separationsavståndet i meter (m).
Fältstyrkor från fasta RF-sändare, somfastställts genom en elektromagnetisk
anläggningsundersökning,a ska vara
mindre än överensstämmelsenivån för
varje frekvensområde.b
Störningar kan inträffa i närheten av
utrustning som är märkt med följandesymbol:OBS 1! Vid 80 MHz och 800 MHz gäller det högre frekvensområdet.
OBS 2! Dessa riktlinjer kanske inte gäller i alla situationer. Elektromagnetisk vågutbredning påverkas av absorption
och reflektion från byggnader, föremål och människor.
aFältstyrkor från fasta sändare, t.ex. basstationer för radiotelefoner (mobila/trådlösa) och fast mobilradio,
amatörradio, AM- och FM-radiosändningar och TV-sändningar, kan inte förutses teoretiskt med noggrannhet.
För att fastställa den elektromagnetiska miljön till följd av fasta RF-sändare bör en elektromagnetisk
anläggningsundersökning övervägas. Om den uppmätta fältstyrkan på den plats där FreeStyle Comfort användsöverstiger den tillämpliga RF-överensstämmelsenivån enligt ovan, ska FreeStyle Comfort hållas under uppsikt för
verifiering av normal drift. Om onormal prestanda observeras kan ytterligare åtgärder bli nödvändiga, såsom att rikta
om eller flytta FreeStyle Comfort.
bÖver frekvensområdet 150 kHz till 80 MHz ska fältstyrkorna vara mindre än 10 V/m.

340 - SV PN MN232-C4 L | Bruksanvisning

FreeStyle Comfort
Rekommenderade separationsavstånd mellan bärbar och mobil
RF-kommunikationsutrustning och FreeStyle Comfort

FreeStyle Comfort är avsedd för användning i en elektromagnetisk miljö där utstrålade RF-störningar kontrolleras.
Kunden eller användaren av FreeStyle Comfort kan hjälpa till att förhindra elektromagnetiska störningar genom
att upprätthålla ett minimiavstånd mellan bärbar och mobil RF-kommunikationsutrustning (sändare) och FreeStyle
Comfort enligt rekommendationerna nedan, i enlighet med kommunikationsutrustningens maximala uteffekt.
Sändarens maximala
Separationsavstånd enligt sändarens frekvens
nominella uteffektm
150 kHz till 80 MHz

80 MHz till 800 MHz

800 MHz till 2,5 GHzW

d = 1.2 P
d = 1.2 P
d = 1.2 P
0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
För sändare med en maximal uteffekt som inte listas ovan kan rekommenderat separationsavstånd (d) i meter (m)beräknas med hjälp av den ekvationen som tillämpas för sändarens frekvens, där P är sändarens maximala uteffekt
i watt (W) enligt sändartillverkaren.
OBS 1! Vid 80 MHz och 800 MHz är det separationsavståndet för det högre frekvensområdet som gäller.
OBS 2! Dessa riktlinjer kanske inte gäller i alla situationer. Elektromagnetisk vågutbredning påverkas av absorption
och reflektion från byggnader, föremål och människor.

Klassificering

Typ av skydd mot elstöt:
Klass II	-skydd mot elektriska stötar uppnås
			genom dubbelisolering.
Grad av skydd mot elektriska stötar:Utrustning av typ BFE
Som tillhandahåller en viss		grad av skydd mot elektriska
		stötar med avseende på		1) läckageström;		2) skyddsjordsanslutningens tillförlitlighet		(om sådan finns).
		Ej avsedd för applicering direkt på hjärtat.
Oberoende tester gällande standarder för elektrisk
utrustning för medicinskt bruk:	Utrustningen har testats för att uppfylla
kraven enligt• IEC 60601-1 utgåva 3.1: Elektrisk utrustning
för medicinskt bruk – del 1: Allmänna
fordringar beträffande säkerhet.
• CAN/CSA C22.2 nr 60601-1-8 utgåva 2.1,
elektrisk utrustning för medicinskt bruk – del 1:
Allmänna fordringar beträffande säkerhet
Skydd mot potentiella elektromagnetiska störningar
eller andra störningar mellan utrustningen och
andra enheter.

	• Utrustningen har testats för att uppfylla kraven
i EN 60601-1-2/IEC 60601-1-2: 4:e utgåvan.
	• Utrustningen har testats för att uppfylla kraven
enligt RTCA/DO160 avsnitt 21 kategori M.
• CISPR 11 / EN 55011 klass B grupp 1,”Industriell, vetenskaplig och medicinteknisk(Industrial, Scientific, and Medical, ISM)
utrustning”• FCC del 15, underavsnitt B – oavsiktligt
utstrålande enheter av klass B
IP22 – Skydd mot intrång av fingrar eller liknande
föremål som är större än 12,5 mm i diameter.
Vertikalt droppande vatten ska inte ha någon
skadlig effekt när höljet lutas i en vinkel på
15° från sin normala position.
Tillåten metod för rengöring och desinfektion:
Se avsnittet ”Rengöring, skötsel och korrekt
underhåll” i den här användarhandboken till
FreeStyle.
Grad av säkerhet vid användning i närheten av
antändliga anestesigaser:	Utrustningen är inte lämplig för sådan
användning.
Driftsätt: Kontinuerlig drift.
PN MN232-C4 L | Bruksanvisning SV- 341

FreeStyle Comfort
Säätimet ja järjestelmän tilan merkkivalot
ISO 7000
Käyttöopas, käyttöohjeet.
Sääntö 1641
Valmistusmaa ja -päivämäärä.
"CC" tarkoittaa valmistusmaan
kaksikirjaimista maakoodia.
Valmistuspäivämäärä on muodossa
VVVV-KK-PP. Sääntö 6049

ISO 7010: Graafiset symbolit–Turvavärit
ja turvamerkit–Rekisteröidyt turvamerkit
Käyttöohjeet on luettava.
Sääntö M002
Pidettävä kaukana avotulesta, tulesta
ja kipinöistä. Avoimet syttymislähteet
ja tupakointi kielletty. Sääntö P003

Säilytettävä kuivassa ja pidettävä
kuivana. Sääntö 0626

Älä tupakoi yksikön lähellä
tai kun käytät sitä. Sääntö P002

Pinoamisraja lukumäärän mukaan,
jossa "n" osoittaa sallittujen yksiköiden
enimmäismäärän. Sääntö 2403

Varoitus. Sääntö W001
IEC 60417
Tyypin BF potilaskosketuksessa
oleva osa (sähköiskusuojauksen taso).
Sääntö 5333

Luettelonumero. Sääntö 2493
Sarjanumero. Sääntö 2498
Säilytys- tai käyttölämpötila-alue.
Sääntö 0632
Säilytystilan kosteusalue. Sääntö 2620
Ilmanpainerajoitus. Sääntö 2621
Valmistajan nimi ja osoite.
Sääntö 3082
Tämä puoli ylös. Sääntö 0623

Luokan II laite, kaksinkertaisesti
eristetty. Sääntö 5172
EU:n asetus 2017/745
Valtuutettu edustaja Euroopan
yhteisössä
Jos tuotteen yksilöllisen
laitetunnisteen (UDI) etiketissäon CE####-symboli, laite täyttää
lääkinnällisiä laitteita koskevat
vaatimukset. CE-symboli ####
ilmaisee ilmoitetun laitoksen
numeron.
Muut symbolit
Pidä kaukana syttyvistä
materiaaleista, öljystä ja rasvasta.

Särkyvää, käsittele varoen.
Sääntö 0621

Ei saa purkaa osiin.

Sisältää vaarallisia aineita.
Sääntö 3723

Lisää virtausta.

Maahantuoja. Sääntö 3725

Vähennä virtausta.

O2-lähtö

Valikkopainike.
Virtakytkin. Sääntö 5009
Täyttää
RTCA/DO160 -osan
21 luokan M
vaatimukset.

342 - FI PN MN232-C4 L | Käyttöopas

Tämän happirikastimen valmistaja
on määrittänyt, että tämä laite
täyttää kaikki sovellettavat FAA:nhyväksymiskriteerit ja että sitä
voidaan käyttää lentokoneessa.

FreeStyle Comfort
Tämän happirikastimen valmistaja
on määrittänyt, että tämä laite
täyttää kaikki sovellettavat FAA:nhyväksymiskriteerit ja että sitä
voidaan käyttää lentokoneessa.

Neuvoston direktiivi 2012/19/EU
WEEE
Tämä symboli muistuttaa laitteen
omistajaa siitä, että elinkaarensa
lopussa laite on kierrätettävä sähköja elektroniikkalaiteromudirektiivin(WEEE) mukaisesti. Tuotteemme
täyttävät vaarallisia aineita koskevan
RoHS-direktiivin vaatimukset.
Tuotteissamme on vain vähäisiä
jäämiä lyijystä ja muista vaarallisista
aineista.

Kierrätyssymboli
Valtuutettu edustaja Sveitsissä(LISÄÄ)Jos laitteessa on UKCA-merkki
UKCA####, joka ilmaisee ilmoitetun
laitoksen numeron, tämä laite on
UKCA-määräysten mukainen.
21 CFR 801.15
Yhdysvaltojen liittovaltion laki sallii
tämän laitteen myynnin vain lääkärilletai lääkärin määräyksestä.

IEC 60601-1-8
Alhaisen prioriteetin tekninen hälytys
ISO 15223-1
Lääkinnällinen laite. Sääntö 5.7.7

ETL-sertifiointi
Sertifioitu Yhdysvaltain ja Kanadan
markkinoille Yhdysvaltain ja Kanadan
soveltuvien standardien mukaisesti
Sisältää FCC-tunnuksen: WAP2001
FCC:n ilmoitus
Tämä laite saattaa sisältää Johnson Technology-yhtiön CYBLE-022001-00-komponentin,
mukaan lukien antenni 2450AT18B100. Nämä
ovat FCC-sääntöjen osan 15 vaatimusten
mukaisia. Tämä laite täyttää FCC:n julkisen
ilmoituksen DA00-1407 mukaisesti hyväksytyt
modulaarisia lähettimiä koskevat vaatimukset.
Lähettimen toimintaa koskevat seuraavat
ehdot: (1) tämä laite ei saa aiheuttaa haitallisia
häiriöitä ja (2) tämän laitteen on vastaanotettava
kaikki häiriöt, mukaan lukien mahdollisesti
ei-toivottua toimintaa aiheuttavat häiriöt. Tuotesaattaa olla suojattu yhdellä tai useammalla
patentilla Yhdysvalloissa ja muissa maissa.
Alla olevalla verkkosivullamme on lueteltu
kaikki sovellettavat patentit. Pat.:www.caireinc.com/corporate/patents

Yksilöllinen laitetunniste 5.7.10
A S/NZS 4417.1, AS/NZS 4417.2
ACMA-säädöstenmukaisuusmerkintä –ilmaisee ACMA-säädöstenmukaisuuden (LISÄÄ)Langaton tekniikka
Liityntätyypit

SSP, iAP2, GATT

Taajuus2402–2480 MHz

Suurin RF-tehon tuotto+4 dBmToiminta-alue

10 m (Luokka 2)

Suosittelemme, että tätä laitetta käytettäessä sen
etäisyys vartalosta on vähintään 1,1 cm. Tämä eikoske kasvosuojaimia, ilmaputkia tai lisälaitteita.

Laitteen käyttöä koskevat graafiset symbolit,
IEC 60417
Tasavirta. Sääntö 5031
Vaihtovirta. Sääntö 5032
IEC 60601-1
Pisarasuojattu

PN MN232-C4 L | Käyttöopas FI - 343

FreeStyle Comfort
FreeStyle Comfort -happirikastimen purkaminen pakkauksestaan
Tarkista, että pakkaus sisältää
kaikki luetellut osat. Jos
jotakin puuttuu, ota heti yhteys
happirikastimen toimittajaan.

Vakiovarusteet• Kannettava happirikastin• Kantolaukku• Olkahihna• 8-kennoinen akkupakkaus• Verkkovirtalähde• Virtajohto• Tasavirtalähde• Käyttöopas

Valinnaiset varusteet• 8-kennoinen lisäakkupakkaus• 16-kennoinen akkupakkaus• Pöytälaturi• Kannettava tarvikelaukku

Tutustuminen FreeStyle Comfort -happirikastimeen
Tutustu FreeStyle Comfortin tärkeimpiin ominaisuuksiin ja sen käyttöliittymään.

Laite edestä
Hihnakiinnitys

Käyttöliittymä

Happilähtö
Poistoilmaaukko

Ilmanotto/
suodattimet
Ilmanotto/
suodattimet

Virtaliitäntä

Akkupakkaus
Akun
vapautuspainike
344 - FI PN MN232-C4 L | Käyttöopas

FreeStyle Comfort
Laite päältä
Näyttöruutu
Lisää virtausta

Virtapainike

Valikkopainike

Vähennä virtausta

Akun näyttöruutu
Virtausasetus

Akun
varausnäyttö

autoSAT-merkki
Jäljellä olevan
akun varaus
prosentteina

Hälytysnäyttö
Bolussyötön
merkkivalo

Sähkövirran näyttöruutu
Virtausasetus
Akun
varausnäyttö

Tila hälytysnäytölle

Latauksenmerkkivalo

PN MN232-C4 L | Käyttöopas FI - 345

FreeStyle Comfort
Happirikastimen toiminta
Hengittämämme ilma sisältää noin 21 %happea, 78 % typpeä ja 1 % muita kaasuja.
FreeStyle® Comfort® -happirikastimeen imetään
huoneilmaa ilmanottoaukkojen kautta. Ilma kulkee
molekyylisiiviläksi kutsutun adsorboivan aineen
läpi. Vain happi läpäisee sen, joten typpi suodattuu
pois. Siksi käyttäjä saa hengitettäväkseen puhdasta
happea.
Tehokas ja helposti mukana kuljetettava FreeStyle
Comfort -happirikastin käyttää kehittynyttä happea
säästävää tekniikkaa ja mahdollistaa kotihoidon.
Edistyksellinen virtaustoiminto tunnistaa, koska
käyttäjä hengittää sisään ja tuottaa happisykäyksen
kunkin sisäänhengitysjakson alkaessa. Tästä
toiminnosta on hyötyä harrastuksissa ja kodin
ulkopuolella liikuttaessa. Sen ansiosta happea ja
akun varausta kuluu vähemmän kuin jatkuvaa
virtausta käytettäessä. FreeStyle Comfort on kevyt,
mikä mahdollistaa sen kuljettamisen olkahihnassa.
Laitetta voi käyttää verkkovirralla (seinäpistoke),ajoneuvosta saatavalla tasavirralla tai akun avulla.

Käyttötarkoitus
CAIRE FreeStyle Comfort -happirikastin ontarkoitettu lisähapen antamiseen. Tätä laitetta ei ole
tarkoitettu elämää ylläpitäviin sovelluksiin, eikä
siinä ole potilaan monitorointiominaisuutta.

Lääkärin määräämä happihoito
Monet kärsivät erilaisista sydän-, keuhko- ja
hengitystiesairauksista. Monille heistä on runsaasti
hyötyä happihoidosta kotona, matkustettaessa tai
poissa kotoa.
Happi on kaasua, jota hengitysilmastammeon 21 %. Kehomme oikea toiminta riippuu
tasaisesta hapensaannista. Lääkärisi on määrännyt
happihoitoa hengitystiesairautesi hoitoon.

346 - FI PN MN232-C4 L | Käyttöopas

Vaikka happihoito ei aiheuta riippuvuutta, se voi
olla vaarallista itsehoitona. Ota yhteys lääkäriin
ennen tämän happirikastimen käyttämistä.
Happirikastimen jälleenmyyjä esittelee, kuinka
lääkärin määräämä jatkuva syöttö säädetään.

Käyttäjät
Happirikastimen avulla syötetään täydentävää
happea käyttäjille, joiden keuhkot eivät pysty
siirtämään sitä riittävästi verenkiertoon.
Kannettavissa happirikastimissa ei ole
happisäiliötä. Niitä ei tarvitse täyttää, ja niiden akut
voidaan ladata verkkovirralla tai tasavirralla missä
vain (lentokonetta lukuun ottamatta). Siksi niitä
voidaan käyttää helposti kotihoidossa tai liikkeellä
oltaessa. Happirikastimen käyttäminen edellyttää
lääkärin määräystä.
Tyypillinen happihoidon käyttäjä on yli 65-vuotias,
jolla on jokin keuhkosairaus, kuten keuhkoahtauma.
Käyttäjillä on yleensä hyvät kognitiiviset kyvyt, ja
heidän täytyy pystyä ilmoittamaan, jos olo muuttuu
huonoksi. Jos käyttäjä ei pysty ilmoittamaan
olon muuttumisesta huonoksi tai lukemaan
ja ymmärtämään happirikastimen merkintöjä
ja käyttöohjetta, on suositeltavaa käyttää sitä
valvotusti. Jos happirikastimen käytön aikana
ilmenee epämukavuutta, käyttäjän tulee ottaa
yhteys terveydenhoidon ammattilaiseen. Lisäksi on
suositeltavaa, että käyttäjällä on varahappijärjestelmä(esimerkiksi happipullo) sähkökatkon ja
happirikastimen vian varalta. Happirikastimen
käyttäminen ei vaadi erityisiä taitoja.

FreeStyle Comfort
Tärkeää!
Turvaohjeet on merkitty seuraavasti:
VAROITUS: TÄRKEITÄ TURVATIETOJAVAAROISTA, JOTKA VOIVATAIHEUTTAA VAKAVIA VAMMOJA.
TÄRKEÄ HUOMAUTUS: tärkeää
tietoa FreeStyle Comfortin vaurioiden
estämiseksi.
Huomautus: nämä tiedot vaativat erityistä huomiota.

Käyttöindikaatiot
VAROITUS: ON TÄRKEÄÄ, ETTÄNOUDATAT LÄÄKÄRIN SINULLE
MÄÄRÄÄMÄÄ HAPPITASOA. ÄLÄ
MUUTA VIRTAUSASETUSTA ILMAN
LÄÄKÄRIN MÄÄRÄYSTÄ.
VAROITUS: TÄTÄ LAITETTA EI OLE TESTATTUTAI HYVÄKSYTTY KÄYTETTÄVÄKSILAPSIPOTILAIDEN HOIDOSSA.
VAROITUS: YHDYSVALTAIN LIITTOVALTIONLAIN MUKAISESTI TÄTÄ LAITETTA SAAOSTAA TAI VUOKRATA VAIN LÄÄKÄRIN TAIMUUN LISENSOIDUN TERVEYDENHUOLLONPALVELUNTARJOAJAN MÄÄRÄYKSESTÄ.
VAROITUS: TÄTÄ YKSIKKÖÄ EI OLE
TARKOITETTU ELÄMÄÄ YLLÄPITÄVIIN
SOVELLUKSIIN. POTILAS, JOKA EI PYSTYILMAISEMAAN LAITTEEN KÄYTÖN AIHEUTTAMAAEPÄMUKAVAA OLOA, SAATTAA TARVITAYLIMÄÄRÄISTÄ TARKKAILUA. KUULO- JA/TAINÄKÖVAMMAISET KÄYTTÄJÄT VOIVAT TARVITAAPUA HÄLYTYSTEN TARKKAILUSSA.
VAROITUS: PULSSIANNOSTELUTILAN ASETUS
ON MÄÄRITETTÄVÄ YKSILÖLLISESTI KUNKIN
POTILAAN TARPEIDEN MUKAISESTI LEVOSSA,LIIKUNNAN AIKANA JA MATKALLA.

VAROITUS: POTILAAN TIETYNLAINEN HENGITYS
EI EHKÄ LAUKAISE HAPPEA SÄÄSTÄVÄÄTOIMINTOA.
VAROITUS: FREESTYLE COMFORT -HAPPIRIKASTINTA EI OLE TARKOITETTU TRAKEOSTOOMAPOTILAAN KÄYTTÖÖN.
VAROITUS: RASKAANA OLEVIEN TAI IMETTÄVIEN
EI PIDÄ KÄYTTÄÄ TÄMÄN KÄYTTÖOHJEENSUOSITTELEMIA VARUSTEITA, SILLÄ NE
SAATTAVAT SISÄLTÄÄ FTALAATTEJA.

Käytön vasta-aiheet
VAROITUS: JOISSAKIN OLOSUHTEISSAVOI OLLA VAARALLISTA ANNOSTELLALISÄHAPPEA ILMAN LÄÄKÄRIN
MÄÄRÄYSTÄ. TÄTÄ LAITETTATULEE KÄYTTÄÄ VAIN LÄÄKÄRIN
MÄÄRÄYKSESTÄ.
VAROITUS: TÄTÄ LAITETTA EI OLE TARKOITETTUKÄYTETTÄVÄKSI MAGNEETTIKUVAUSYMPÄRISTÖISSÄ TAI NIIDEN LÄHEISYYDESSÄ, KUTEN
MK-LAITTEISSA.
VAROITUS: EI SAA KÄYTTÄÄ HELPOSTISYTTYVIEN ANESTESIA-AINEIDEN LÄHELLÄ.
VAROITUS: KAIKKIEN SÄHKÖLAITTEIDEN TAVOIN
TÄMÄ HAPPIRIKASTIN EI OLE KÄYTETTÄVISSÄ,JOS SÄHKÖNSYÖTTÖ SIIHEN KATKEAA TAIPÄTEVÄ HUOLTOTEKNIKKO HUOLTAA SITÄ.
TÄMÄ HAPPIRIKASTIN EI SOVI KÄYTTÄJILLE,
JOIDEN TERVEYTEEN TILAPÄINEN KATKOSVAIKUTTAA HAITALLISESTI.
VAROITUS: TÄTÄ LAITETTA EI OLE HYVÄKSYTTYKÄYTETTÄVÄKSI KAIKISSA YMPÄRISTÖISSÄ.
ALTISTUMINEN TIETYILLE SÄHKÖMAGNEETTISENSÄTEILYN LÄHTEILLE VOI AIHEUTTAA HÄIRIÖITÄ
TAI LAITTEEN TAHATONTA TOIMINTAA. HÄIRIÖNSATTUESSA LAITE ON SAMMUTETTAVAVÄLITTÖMÄSTI.

VAROITUS: TÄMÄ LAITE ON TARKOITETTUKÄYTETTÄVÄKSI KOTIHOIDOSSA JA
VASTAAVISSA AVOHOITOYMPÄRISTÖISSÄ.
VAROITUS: TÄMÄ LAITE ON TARKOITETTU
HAPPIHOITOA SAAVAN POTILAAN KÄYTTÖÖN.

PN MN232-C4 L | Käyttöopas FI - 347

FreeStyle Comfort
Turvallisuusohjeet
VAROITUS: TÄTÄ LAITETTA EI SAA
MUOKATA.
VAROITUS: LAITE ANNOSTELEE ERITTÄIN
HAPPIPITOISTA ILMAA, JOKA KIIHDYTTÄÄ
PALAMISTA. TUPAKOINTI JA AVOTULENKÄYTTÄMINEN ON KIELLETTYÄ SAMASSAHUONEESSA (1) TÄMÄN LAITTEEN KANSSA TAI(2) MUUN HAPPILISÄVARUSTEEN KANSSA. TÄMÄN
VAROITUKSEN LAIMINLYÖMINEN VOI AIHEUTTAAVAKAVAN TULIPALON, OMAISUUSVAHINKOJA,FYYSISIÄ VAMMOJA TAI KUOLEMAN.
VAROITUS: ÄLÄ KÄYTÄ LAITETTA AHTAASSATAI SULJETUSSA TILASSA (KUTEN PIENESSÄLAATIKOSSA TAI KÄSILAUKUSSA), JOSSAILMANVAIHTO ON HEIKKO. SE VOI AIHEUTTAA
HAPPIRIKASTIMEN YLIKUUMENEMISEN JA
TEHON HEIKKENEMISEN.
VAROITUS: HAPPIRIKASTINTA EI SAA ALTISTAASAVULLE, ILMANSAASTEILLE EIKÄ HÖYRYILLE.
VAROITUS: JOS TÄMÄN LAITTEEN KANSSAILMENEE VAKAVA VAARATILANNE, KÄYTTÄJÄN
TULEE VÄLITTÖMÄSTI ILMOITTAA SIITÄ
MYYJÄLLE JA/TAI VALMISTAJALLE. VAKAVALLA
VAARATILANTEELLA TARKOITETAANLOUKKAANTUMISTA, KUOLEMAA TAIMAHDOLLISTA LOUKKAANTUMISTA/KUOLEMAA,JOS VAARATILANNE TOISTUU. KÄYTTÄJÄ VOI
MYÖS ILMOITTAA VAARATILANTEESTA SEN
MAAN TOIMIVALTAISELLE VIRANOMAISELLE,JOSSA VAARATILANNE TAPAHTUI.
VAROITUS: JOS KÄYTETÄÄN
HAPPILISÄVARUSTEITA, JOITAEI OLE SUOSITELTU TÄLLE
HAPPIRIKASTIMELLE, LAITTEENTEHO VOI HEIKENTYÄ. SUOSITELLUTLISÄVARUSTEET ON MAINITTU TÄSSÄ
KÄYTTÖOPPAASSA.
VAROITUS: JOS HAPPIRIKASTIN ON PUDONNUT,
VAHINGOITTUNUT TAI ALTISTUNUT VEDELLE,OTA YHTEYTTÄ LAITTEEN TOIMITTAJAAN,
JOTTA LAITE TARKISTETAAN JA TARVITTAESSA
KORJATAAN. ÄLÄ KÄYTÄ HAPPIRIKASTINTA,JOS SEN VIRTAJOHTO TAI PISTOKE ON
VAHINGOITTUNUT.
VAROITUS: HAPPIRIKASTIMENVAHINGOITTUMISEN TAI HENKILÖVAMMAN
VÄLTTÄMISEKSI KIINNITÄ SE TIUKASTIESIMERKIKSI VYÖLLÄ, KUN KULJETAT SITÄAJONEUVOSSA.
348 - FI PN MN232-C4 L | Käyttöopas

VAROITUS: YHDYSVALTAIN LIIKENNEMINISTERIÖN(DOT) JA YHDISTYNEIDEN KANSAKUNTIEN (YK)SÄÄDÖSTEN MUKAISESTI LAITTEEN AKKU ON
POISTETTAVA KAIKILLA KANSAINVÄLISILLÄLENNOILLA, KUN HAPPIRIKASTIN KIRJATAAN
MATKATAVARAKSI. KUN HAPPIRIKASTIN
TOIMITETAAN POSTITSE, AKKU ON POISTETTAVALAITTEESTA JA PAKATTAVA ASIANMUKAISESTI.
VAROITUS: ÄLÄ KÄYTÄ TÄTÄ LAITETTA, ENNEN
KUIN OLET LUKENUT TÄMÄN KÄYTTÖOPPAAN
JA YMMÄRTÄNYT, MITÄ SIINÄ SANOTAAN.
JOS ET YMMÄRRÄ VAROITUKSIA JA OHJEITA,OTA YHTEYTTÄ LAITETOIMITTAJAANENNEN LAITTEEN KÄYTTÄMISTÄ. MUUTOINSEURAUKSENA VOI OLLA HENKILÖVAHINKOTAI LAITEVAURIO.
VAROITUS: JOS OLOSI ON EPÄMUKAVA TAIKOHTAAT LÄÄKETIETEELLISEN HÄTÄTILAN,
HAKEUDU VÄLITTÖMÄSTI LÄÄKÄRINHOITOON.
VAROITUS: JOS HAPPIRIKASTINTA KÄYTETÄÄNSUOSITELTUJEN KÄYTTÖLÄMPÖTILOJENULKOPUOLELLA, HAPPIRIKASTIMEN TOIMINTA
EI EHKÄ VASTAA TEKNISISSÄ TIEDOISSAILMOITETTUA. TARKISTA LÄMPÖTILAT TÄMÄN
KÄYTTÖOPPAAN TEKNISISTÄ TIEDOISTA.
VAROITUS: ÄLÄ KÄYTÄ ÖLJYÄ, RASVAA TAIMAAÖLJYPOHJAISIA TAI MUITA HELPOSTISYTTYVIÄ TUOTTEITA HAPPILISÄVARUSTEIDENTAI HAPPIRIKASTIMEN KANSSA. KÄYTÄ VAIN
VESIPOHJAISIA, HAPEN KANSSA YHTEENSOPIVIAKOSTEUSEMULSIOITA TAI VOITEITA. HAPPIKIIHDYTTÄÄ HELPOSTI SYTTYVIEN AINEIDEN
PALAMISTA.
VAROITUS: JOS AKKUA KÄYTETÄÄN VÄÄRIN,AKKU VOI KUUMETA TAI SYTTYÄ PALAMAAN,
MISTÄ VOI SEURATA VAKAVA VAMMA.
AKKUA EI SAA PUHKAISTA, LYÖDÄ, TALLOA
EIKÄ PUDOTTAA, EIKÄ SITÄ SAA ALTISTAAVOIMAKKAILLE ISKUILLE EIKÄ TÄRÄHDYKSILLE.
VAHINGOITTUNEEN AKUN KÄYTTÄMINEN VOIAIHEUTTAA HENKILÖVAHINKOJA.
VAROITUS: ÄLÄ OIKOSULJE AKUN NAPOJAYHDISTÄMÄLLÄ NE TOISIINSA AVAINTEN TAIKOLIKOIDEN KALTAISILLA METALLIESINEILLÄ.
MUUTOIN VOI AIHEUTUA KIPINÖINTIÄ TAIVOIMAKASTA KUUMENEMISTA.

FreeStyle Comfort
VAROITUS: KUN KANNETTAVAA HAPPIRIKASTINTAKÄYTETÄÄN ULKONA VAIHTOVIRTALÄHTEENKANSSA, KYTKE VIRTALÄHDE AINOASTAANVIKAVIRTAKATKAISIMELLA VARUSTETTUUNPISTORASIAAN.
VAROITUS: HAPPIRIKASTINTA EI SAAKÄYTTÄÄ MUIDEN LAITTEIDEN VIERESSÄ TAIPINOTTUNA MUIDEN LAITTEIDEN KANSSA. JOSLAITETTA JOUDUTAAN KÄYTTÄMÄÄN MUIDENLAITTEIDEN VIERESSÄ TAI PINOTTUNA NIIDENKANSSA, LAITETTA PITÄÄ VALVOA NORMAALIN
TOIMINNAN VARMISTAMISEKSI.
VAROITUS: MUIDEN KUIN MÄÄRITETTYJEN
KAAPELEIDEN TAI SOVITTIMIEN KÄYTTÄMINEN,LUKUUN OTTAMATTA LÄÄKINNÄLLISEN
SÄHKÖLAITTEEN VALMISTAJAN TOIMITTAMIA
KAAPELEITA JA SOVITTIMIA, JOTKA ON
TARKOITETTU SISÄISTEN OSIEN VARAOSIKSI,VOI LISÄTÄ HAPPIRIKASTIMEN PÄÄSTÖJÄ TAIVÄHENTÄÄ SEN SUOJAUSTA.
VAROITUS: KÄYTÄ VAIN SITÄ SÄHKÖJÄNNITETTÄ,JOKA ON MÄÄRITETTY LAITTEESEENKIINNITETYSSÄ TEKNISTEN TIETOJEN TARRASSA.
VAROITUS: ÄLÄ KÄYTÄ JATKOJOHTOJA TÄMÄN
LAITTEEN KANSSA ÄLÄKÄ KYTKE LIIKAASÄHKÖLAITTEITA SAMAAN PISTORASIAAN.
JATKOJOHTOJEN KÄYTTÄMINEN VOIHEIKENTÄÄ LAITTEEN TEHOA HUOMATTAVASTI.
JOS YHTEEN PISTORASIAAN KYTKETÄÄN
LIIKAA SÄHKÖLAITTEITA, SÄHKÖTAULUVOI KUORMITTUA, MIKÄ VOI LAUKAISTAKATKAISIMEN/SULAKKEEN TAI TULIPALOON,JOS KATKAISIN/SULAKE EI TOIMI.
TÄRKEÄ HUOMAUTUS: Jos laiteantaa varoituksen tai jos havaitset,
että happirikastin ei toimi oikein,katso lisätietoja tämän käyttöoppaan
vianetsintäosiosta. Jos ongelma ei
ratkea, ota yhteyttä laitetoimittajaan.

VAROITUS: ÄLÄ KOHDISTALAITTEESEEN MITÄÄN HUOLTOTOIMIALUKUUN OTTAMATTA TÄSSÄ
KÄYTTÖOPPAASSA MAINITTUJARATKAISUJA.
VAROITUS: KÄYTÄ VAIN VALMISTAJANSUOSITTELEMIA LISÄVARUSTEITA. MUIDEN
LISÄVARUSTEIDEN KÄYTTÄMISESTÄ VOIAIHEUTUA VAARAA, HAPPIRIKASTIN VOIVAHINGOITTUA VAKAVASTI TAI TAKUU VOIKUMOUTUA.
VAROITUS: VALMISTAJA SUOSITTELEE
VARAHAPPIJÄRJESTELMÄÄ SÄHKÖKATKON,LAITTEEN HÄLYTYSTILAN TAI MEKAANISEN VIAN
VARALTA. PYYDÄ LISÄTIETOJA TARVITTAVASTAVARAJÄRJESTELMÄSTÄ LÄÄKÄRILTÄSI TAILAITETOIMITTAJALTASI.
VAROITUS: ÄLÄ TUKI TULO- TAI POISTOILMAAUKKOJA. ÄLÄ PUDOTA TAI ASETAMITÄÄN ESINEITÄ LAITTEEN AUKKOIHIN.
SE VOI AIHEUTTAA HAPPIRIKASTIMENYLIKUUMENEMISEN JA TEHON HEIKKENEMISEN.
VAROITUS: TUPAKOINTI HAPPILAITTEENKÄYTTÄMISEN YHTEYDESSÄ ON YLEISIN SYY
TULIPALOIHIN JA NIIHIN LIITTYVIIN KUOLEMIIN.
NOUDATA NÄITÄ TURVAVAROITUKSIA:
VAROITUS: TUPAKOINTI, KYNTTILÖIDEN
POLTTAMINEN JA AVOTULEN KÄYTTÄMINEN ON
KIELLETTYÄ SAMASSA HUONEESSA TÄMÄN
LAITTEEN TAI HAPPILISÄVARUSTEIDEN KANSSA.
VAROITUS: TUPAKOINTI HAPPIKANYYLIN
KÄYTTÄMISEN AIKANA VOI AIHEUTTAAPALOVAMMOJA KASVOILLE JA MAHDOLLISESTIKUOLEMAN.
VAROITUS: JOS POLTAT, NÄMÄ 3 VAIHETTAVOIVAT PELASTAA HENKESI: SAMMUTA
HAPPIRIKASTIN, IRROTA KANYYLI JA POISTUHUONEESTA, JOSSA LAITE ON.

TÄRKEÄ HUOMAUTUS: Noudata valmistajan
ohjeita, jotta takuu ei kumoudu.

PN MN232-C4 L | Käyttöopas FI - 349

FreeStyle Comfort
VAROITUS: "TUPAKOINTI KIELLETTY – HAPPILAITE
KÄYTÖSSÄ" -KYLTIN ON OLTAVA NÄKYVÄSTI
ESILLÄ KOTONA TAI MUUSSA TILASSA, MISSÄ
HAPPILAITETTA KÄYTETÄÄN. POTILAILLE
JA HEIDÄN HOITAJILLEEN ON KERROTTAVATUPAKOINNIN VAAROISTA LÄÄKINNÄLLISEN
HAPEN LÄHEISYYDESSÄ JA SITÄ KÄYTETTÄESSÄ.
VAROITUS: ÄLÄ KÄYTÄ HAPPIRIKASTINTAHELPOSTI SYTTYVIEN KAASUJEN LÄHELLÄ.
SE VOI AIHEUTTAA NOPEAN PALON, MISTÄ VOI
OLLA SEURAUKSENA OMAISUUSVAHINKOJA,FYYSISIÄ VAMMOJA TAI KUOLEMA.
VAROITUS: IRROTETUN KANYYLIN ASETTAMINENVAATTEIDEN, VUODEVAATTEIDEN, SOHVANTAI MUUN TEKSTIILIMATERIAALIN PÄÄLLE VOIAIHEUTTAA RÄJÄHTÄVÄN PALON, JOS KANYYLIALTISTUU SAVUKKEELLE, LÄMMÖNLÄHTEELLE
TAI AVOTULELLE.
VAROITUS: ÄLÄ JÄTÄ NENÄKANYYLIAVAATTEIDEN, VUODEVAATTEIDEN TAIISTUINTYYNYJEN PÄÄLLE TAI ALLE. JOSLAITTEEN VIRTA ON KYTKETTY MUTTA SE EI
OLE KÄYTÖSSÄ, HAPPI TEKEE MATERIAALISTAHELPOSTI SYTTYVÄN. KUN HAPPIRIKASTIN EI
OLE KÄYTÖSSÄ, KATKAISE SIITÄ VIRTA.
VAROITUS: HYVÄKSYTYT HAPEN
ANNOSTELUSSA KÄYTETTÄVÄT TARVIKKEET
ON SUUNNITELTU VÄHENTÄMÄÄN PALON
LEVIÄMISTÄ POTILAAN TURVALLISUUDENVUOKSI. HYVÄKSYMÄTTÖMIEN
LISÄVARUSTEIDEN KÄYTTÖ VOI LISÄTÄ
TULIPALOIHIN LIITTYVIEN VAMMOJEN TAI
VAURIOIDEN RISKIÄ.
VAROITUS: HAPPIRIKASTIMEN VIRTA ON
KATKAISTAVA JA SE ON IRROTETTAVAVERKKOVIRRASTA ENNEN AKKUPAKKAUKSEN
IRROTTAMISTA.
VAROITUS: PIENET OSAT VOIVATMUODOSTAA TUKEHTUMISVAARAN.
VAROITUS: ÄLÄ KOSKE POISTOILMA-AUKKOONYLI 10 SEKUNTIA PIDEMPÄÄ AIKAA, SILLÄ SE
SAATTAA OLLA LÄMMIN.
VAROITUS: TUULI TAI VOIMAKAS VETO SAATTAAHAITATA HAPPIHOITOA.
VAROITUS: TÄMÄ TUOTE VOI ALTISTAA SINUTKEMIKAALEILLE, MUKAAN LUKIEN NIKKELI, JOKA
KALIFORNIAN OSAVALTION TIETOJEN MUKAANAIHEUTTAA SYÖPÄÄ. KATSO LISÄTIETOJASIVULTA WWW.P65WARNINGS.CA.GOV.

350 - FI PN MN232-C4 L | Käyttöopas

TÄRKEÄ HUOMAUTUS: irrota
vaihtovirtalähde seinäpistorasiasta aina
ennen kuin irrotat vaihtovirtalähteen
happirikastimesta.
TÄRKEÄ HUOMAUTUS: älä sijoita happirikastinta
niin, että virtajohtoon on vaikea päästä käsiksi.
TÄRKEÄ HUOMAUTUS: aseta hapen syöttöputket
ja virtajohdot siten, että niihin ei voi kompastua
eivätkä ne aiheuta kuristumisen vaaraa.
TÄRKEÄ HUOMAUTUS: Kun käytät
happirikastinta auton, veneen tai muun ajoneuvon
tasavirtalähteellä, varmista, että ajoneuvo on
käynnistetty ja käynnissä, ennen kuin kytket
happirikastimen. Jos näitä ohjeita ei noudateta,
seurauksena voi olla, että virtalähde ei anna virtaa
happirikastimeen.
TÄRKEÄ HUOMAUTUS: Kun auto, jossa käytät
happirikastinta, ei ole käynnissä, irrota ja poista laite
autosta. Älä säilytä happirikastinta hyvin kuumassa
tai kylmässä autossa tai vastaavassa ympäristössä,jonka lämpötila on hyvin korkea tai matala.
Huomautus: kannettavat ja siirrettävät radiotaajuiset
viestintävälineet voivat häiritä sähkökäyttöisiä
lääkintälaitteita.
TÄRKEÄ HUOMAUTUS: ÄLÄ pidä
happirikastinta tai virtalähdettä kytkettynä
ajoneuvoon, jos auton virta on katkaistu.
Tämä voi tyhjentää ajoneuvon akun.
TÄRKEÄ HUOMAUTUS: Jos happirikastinta käytetään
tai säilytetään sen normaalien käyttölämpötilarajojen
ja painealueen (korkeus) ulkopuolella, sensuorituskyky voi heikentyä, akun käyttöaika voi
lyhentyä ja/tai akun lataamiseen kuluva aika voi
pidentyä. Tarkista säilytys- ja käyttölämpötilarajat
tämän käyttöoppaan teknisistä tiedoista.
TÄRKEÄ HUOMAUTUS: Jos happirikastimessa
on akku: Säilytä viileässä ja kuivassa paikassa,
jotta akun käyttöikä ei lyhene. Jos happirikastinta
säilytetään pitkään korkeassa lämpötilassa tai
täyteen ladatun tai täysin tyhjentyneen akun
kanssa, akun käyttöikä voi lyhentyä. Älä yritä avata
akkua. Se ei sisällä huollettavia osia. Pidä akut
lasten ulottumattomissa.
TÄRKEÄ HUOMAUTUS: Käytä VAIN valmistajan
toimittamia akkuja. Hävitä akku laitetoimittajan
tai paikallisten viranomaisten kierrätysohjeiden
mukaisesti.

FreeStyle Comfort
Huomautus: Rikastimen häiriön tai sähkökatkon
aiheuttamien haittojen lieventämiseksi laitetoimittajan
on arvioitava varahapen tarve asennuksen aikana ja
ajoittain, kun seuraavat seikat muuttuvat:– potilaan tila
– potilaan asuinympäristö– mahdollisuus antaa potilaalle lisähappea
varahappilähteestä.
Huomautus: Varmista, että kanyyli on asetettu
kokonaan ja pitävästi. Tämä varmistaa, että
happirikastin voi tunnistaa hengityksen oikein hapen
annostelua varten. Sisäänhengityksen aikana kuulet
tai tunnet, kuinka happi virtaa nenäkanyylin haaroihin.

Huomautus: Jos happirikastinta on säilytetty pitkään
sen normaalien käyttölämpötilarajojen ulkopuolella(ne on kuvattu teknisten tietojen taulukossa), sen
on annettava palautua normaaliin käyttölämpötilaan
ennen sen käynnistämistä.
Huomautus: Laitetoimittajan on arvioitava rikastimen
häiriön tai sähkökatkon takia tarvittavan varahapen
tarve asennuksen aikana ja ajoittain, kun seuraavat
seikat muuttuvat:– potilaan tila
– potilaan asuinympäristö– mahdollisuus antaa potilaalle lisähappea
varahappilähteestä.

Huomautus: Jos litiumakut altistuvat erittäin kuumalle
lämpötilalle, kun ne on ladattu täyteen tai ne ovat
täysin tyhjiä, ne voivat menettää kapasiteettinsa
pysyvästi. Ennen akkujen varastoimista pitkään on
suositeltavaa ladata niitä 25–50 %. Suositeltava
säilytyslämpötila on –20...20 °C (–4...68 °F).
Huomautus: Älä käytä happirikastinta, jostuloilmasuodatin ei ole paikallaan. Jos mukana on
toinen suodatin, aseta varasuodatin, ennen kuin
puhdistat likaisen suodattimen. Puhdista likainen
suodatin lämpimällä saippuavesiliuoksella ja anna sen
kuivua kokonaan ennen käyttöä.
Huomautus: Valmistaja ei suosittele tämän laitteen
sterilisointia.
Huomautus: Hapen loppumisen vaaraa ei ole
huoneessa, jossa happirikastinta käytetään.
VAROITUS: HAPPIRIKASTINTA EI SAAKÄYTTÄÄ TAI LIITTÄÄ ULKOISEENTASA- TAI VAIHTOVIRTALÄHTEESEENILMAN AKKUA. JOS LAITTEESTAON IRROTETTU AKKU, ÄLÄ YRITÄ
KÄYNNISTÄÄ SITÄ TAI YHDISTÄÄ SITÄULKOISEEN VIRTALÄHTEESEEN.
IRROTA AKKU VASTA KUN LAITTEESTA ON
KATKAISTU VIRTA JA SE ON IRROTETTU
VIRRANSYÖTÖSTÄ. ÄLÄ YRITÄ POISTAA AKKUA,KUN LAITTEESEEN ON KYTKETTY VIRTA TAI SE
ON YHDISTETTYNÄ VIRTALÄHTEESEEN.

PN MN232-C4 L | Käyttöopas FI - 351

FreeStyle Comfort
Tekniset tiedot
FreeStyle Comfort
Mitat (yksi akku)

25,4 x 18,5 x 7,9 cm (10,0 x 7,3 x 3,1 tuumaa)

Mitat (kaksi akkua)

27,9 x 18,5 x 7,9 cm (11,0 x 7,3 x 3,1 tuumaa)

Paino (yksi akku)

2,3 kg (5 paunaa)

Paino (kaksi akkua)

2,7 kg (6 paunaa)

Nimellisäänitaso***

39,93 dB(A) ± 0,42 dB(A) @ 2

Virtausasetukset
Hapen ulostuloasetus([@ 20 bpm STPD] Kaikki tilavuudet +/-15 %)Happipitoisuus*
Happimittari

Sykäysasetus 1, 2, 3, 4, 5
Asetus 1–210 ml/min
Asetus 2–420 ml/min
Asetus 3–630 ml/min
Asetus 4–840 ml/min
Asetus 5–1 050 ml/min
90 % (+5,5 % / -3 %)Kyllä

Maksimilähtöpaine< 30 PSI

Annostelutarkkuus> –0,1 cm H2O (säädettävissä)

vaihtovirta

100–240 VAC, 50–60 Hz, 120 W enint.

tasavirta

11–18 VDC (10 A maks.)

Akkutyyppi
Akun kapasiteetti
Akun uudelleenlatausaika
Akun kesto
Käyttölämpötila**

Litiumioniakku
Yksikennoinen: 6 700 mAh; kaksikennoinen: 13 400 mAh
Yksi akku: 3,5 tuntia
Kaksi akkua: 6,0 tuntia
Yhden akun kesto: Asetus 2 – jopa 4 tuntia
Kahden akun kesto: Asetus 2 – jopa 8 tuntia
5–40 °C (41–104 °F)

Kuljetus- ja varastointilämpötila (laite)–25...70 °C (–13...+158 °F), maks. 3 048 m (10 000 ft)

Kuljetus- ja varastointilämpötila (akku)–25...70 °C (–13...+158 °F), maks. 3 048 m (10 000 ft)

Käyttökorkeus**
Käyttöympäristön kosteus
Säilytystilan kosteus
Rajoitettu takuu–381...+3 048 m (–1 250–10 000 ft) (testattu 700–1 060 hPa:nilmanpaineelle)
15–95 % suhteellinen kosteus (ei kondensoiva)
0–90 %, kondensoitumaton
Katso takuuilmoitus* Perustuu ilmanpaineeseen 14,7 psi (101 kPa) lämpötilassa 21 °C (70 °F)** Laite toimii kaikissa ympäristöissä, kunhan ympäristön lämpötila ja ilmankosteus ovat määritetyillä alueilla.
Jos happirikastinta käytetään muunlaisissa olosuhteissa, se ei ehkä toimi teknisten tietojensa mukaisesti. Lämmitysnäyttö
saattaa jäädä näkyviin, kunnes happirikastin on saavuttanut tavoitepitoisuuden tai 15 minuutiksi. Jatka laitteen käyttöä
kyseisenä aikana.
*** Äänitaso on mitattu testausmenetelmän nro 14-1 10/2018 MDS-Hi mukaisesti.
Odotettavissa oleva käyttöikä on vähintään viisi vuotta.

352 - FI PN MN232-C4 L | Käyttöopas

FreeStyle Comfort
Käyttöohjeet
1. Sijoita FreeStyle Comfort käytön kannalta
sopivaan paikkaan.
a. Happirikastimen ympärillä tulee olla hyvä
ilmankierto, jotta ilmanottosuodattimet
ja ilmanpoistoaukot eivät tukkeutuisi.
b. Happirikastin tulee asettaa paikkaan, jossa
merkkivalot näkyvät ja hälytykset kuuluvat
esteettä.
c. Varmista, että ilmanottosuodattimet ovat
paikoillaan ennen laitteen käyttämistä.
Jos laitteesta puuttuu suodatin, ota yhteyttä
laitteen toimittajaan.
2. Asenna akkupakkaus happirikastimeen.
a. Kohdista akku FreeStyle Comfort-happirikastimen keskelle kotelon alaosaa.

3. Työnnä virtajohto laitteen etupuolella oikeassaalareunassa olevaan liittimeen.

4. Yhdistä nenäkanyyli hapen syöttöliitäntään.
Varmista ennen käyttämistä, että liitos on tiivis.
HUOMAUTUS: muiden kuin suositeltujen kanyyleiden
käyttö voi heikentää laitteen tehoa.

Huomautus: Laitetoimittajalle: Seuraavia
happilisävarusteita suositellaan käytettäväksi
FreeStyle Comfort -happirikastimen kanssa:b. Työnnä akkua, kunnes kuulet napsahduksen
osoituksena sen lukkiutumisesta paikalleen.

• Nenäkanyyli 2,1-metrisellä (7 jalkaa) letkulla (suurin
LPM: 6 LPM): CAIRE-osanumero 5408-SEQ• Lisätietoja muista suositelluista lisävarusteista on
saatavilla verkossa osoitteessa www.caireinc.com.
5. Käynnistä FreeStyle Comfort painamalla
virtapainiketta ja valitse haluamasi virtausasetus
lisäämällä virtausta (+) tai vähentämällä
virtausta (-).

PN MN232-C4 L | Käyttöopas FI - 353

FreeStyle Comfort

Laite antaa lyhyen äänimerkin, joka osoittaa,
että laite on päällä.
Huomautus: Kun FreeStyle Comfort käynnistetään
ensimmäistä kertaa, virtauksen oletusasetus on 2.
Jos asetusta muutetaan, laite muistaa edellisen
asetuksen uuden käynnistyksen yhteydessä.
6. Aseta nenäkanyyli kasvoillesi noudattaen
sen valmistajan ohjeita. Hengitä normaalisti.
Kun hengität sisään, tunnet, että nenäkanyylistä
virtaa happisykäys.
7. Voit muuttaa virtausasetusta painamalla
lisää virtausta (+) tai vähemmän virtausta (-)
haluamasi virtausmäärän saavuttamiseksi.
Kun virtausasetus vaihtuu, ääni muuttuu.

Hengitysnopeus
15
20
25
30
35
40

autoSAT-sykäystilavuuden (ml)* asetus
1
2
3
4
5
10,5
10,5
10,5
10,5
10,5
10,5

21,0
21,0
21,0
21,0
21,0
21,0

31,5
31,5
31,5
31,5
30,0
26,3

* Toleranssi on +/- 15 %354 - FI PN MN232-C4 L | Käyttöopas

42,0
42,0
42,0
35,0
30,0
26,3

52,5
52,5
42,0
35,0
30,0
26,3

Hengitysnopeus
15
20
25
30
35
40

Ei-autoSAT-sykäystilavuuden (ml)* asetus
1
2
3
4
5
14,0
10,5
8,4
7,0
6,0
5,3

28,0
21,0
16,8
14,0
12,0
10,5

42,0
31,5
25,2
21,0
18,0
15,8

56,0
42,0
33,6
28,0
24,0
21,0

70,0
52,5
42,0
35,0
30,0
26,3

8. Käännä FreeStyle Comfort pois päältä
painamalla virtapainikkeesta.
Huomautus: Näyttöruutu siirtyy valmiustilaan ja
himmenee 20 sekunnin kuluttua. Näyttö herää
valmiustilasta, kun painat mitä tahansa painiketta.
9. Tietoja erikoistoiminnoista on kohdassa Valikonselausohjeet.

TUOTTEEN ERITYISTOIMINNOT
autoSAT®
CAIREn autoSAT-ominaisuuden ansiosta
FreeStyle Comfort annostelee tasaisen
happimäärän (annoksen) hengitystä kohti
enimmäishengitysnopeuteen asti. Kun käyttäjän
hengitysnopeus kasvaa esimerkiksi kävelyn aikana,
autoSAT-toiminto säätää laitteen toimintaa niin,
että asetettu annoksen tilavuus säilyy.
UltraSense®
Herkkä sykäysannoksen käynnistin havaitsee
hengenvedon aikaisen painemuutoksen ja antaa
happisykäyksen sisäänhengityksen aikana.
autoDOSE®
Ominaisuus antaa Hengitystä ei havaittu-hälytyksen yhteydessä happisykäyksiä
20 hengenvetoa minuutissa -nopeudella laitteen
senhetkisen virtausasetuksen mukaisesti, kunnes
hälytystila raukeaa.

FreeStyle Comfort
Virtalähteet
FreeStyle Comfort voi saada virtansa kolmesta eri
lähteestä. Niitä ovat:• ladattava akkupakkaus• vaihtovirta• tasavirta
Toiminnasta kerrotaan seuraavissa luvuissa.
VAROITUS: KÄYTÄ VAIN TÄSSÄ
KÄYTTÖOHJEESSA KUVATTUJAVIRTALÄHTEITÄ. MUIDEN KUIN CAIRENTOIMITTAMIEN VIRTALÄHTEIDEN
JA AKKUJEN KÄYTTÄMINEN VOIVAURIOITTAA LAITETTA JA AIHEUTTAATOIMINTAHÄIRIÖN.

AKUN IRROTTAMINEN

Työnnä akun vapautuspainiketta alaspäin ja
poispäin FreeStyle Comfort -happirikastimesta.
Pidä painiketta painettuna ja irrota akku
happirikastimesta vetämällä.
VAROITUS: HAPPIRIKASTIMENPITÄÄ OLLA POIS PÄÄLTÄ JA
IRTI VERKKOVIRRASTA ENNENAKKUPAKKAUKSEN IRROTTAMISTA.

AKKUPAKKAUS

FreeStyle Comfort -happirikastinta voidaan käyttää
ladattavalla akulla, joka toimitetaan laitteen
mukana. Kun happirikastin saa virtaa akusta,
mitään toimintarajoituksia ei aiheudu. Kaikki
virtausvaihtoehdot ovat tällöin käytettävissä.

AKKUPAKKAUKSEN LATAUKSENTARKISTAMINENAKUN ASENTAMINEN

Kohdista akku FreeStyle Comfort -happirikastimen
keskelle kotelon alaosaa. Työnnä akkua,
kunnes kuulet napsahduksen osoituksena sen
lukkiutumisesta paikalleen.

Voit tarkistaa akun varaustason sen ollessa
asennettuna FreeStyle Comfort -happirikastimeen
akkukuvakkeesta ja jäljellä olevasta käyttöajasta
laitteen näytöltä.
80–100%60–80%

40–60%

20–40%

20%

PN MN232-C4 L | Käyttöopas FI - 355

FreeStyle Comfort
Huomautus: jos virtalähde on liitetty, näytöllä näkyy
pistotulpan kuvake prosenttien sijaan.
Kun akkua ei ole asennettu FreeStyle Comfort-happirikastimeen, voit tarkistaa sen varaustason
asettamalla sen tasaiselle alustalle ja painamalla
TEST-painiketta. TEST-painikkeen oikealla
puolella näkyvät akkumittarin merkkivalot(25–100 %) ilmaisevat akun lataustason.

AKKUKÄYTTÖAIKA

Kun rikastinta ei ole kytketty ulkoiseen tasa- tai
vaihtovirtaan, se siirtyy automaattisesti akkukäytölle,jos laitteeseen on asennettu ladattu akku. Uuden
akun kesto kuvataan alla näkyvässä taulukossa.
Hengitysnopeus ei vaikuta akun kestoon.
Sykäysasetus

Yhden(8-kennoisen)
akun kesto

Kaksoisakun(16-kennoisen)
kesto

1

jopa 8 tuntia

jopa 16 tuntia

2jopa 4 tuntia

jopa 8 tuntia

3

jopa 3 tuntia

jopa 6 tuntia

4

jopa 2,25 tuntia

jopa 4,5 tuntia

5

jopa 2 tuntia

jopa 4 tuntia

TYYPILLINEN AKUN LATAUSAIKALATAAMINEN ENSIMMÄISEN KERRANFreeStyle Comfort -happirikastimen mukana
toimitettu uusi akku on ladattu tehtaalla vain
osittain. Ennen akun käyttämistä ensimmäistä
kertaa lataa se täyteen verkko- tai tasavirran avulla.

YLEISTÄ TIETOA LATAAMISESTAFreeStyle Comfort lataa akkupakkauksen aina, kun
se on asennettuna rikastimeen ja kun rikastin on
kytkettynä ulkoiseen vaihto- tai tasavirtalähteeseen(lentokonetta lukuun ottamatta).
Akku on täyteen ladattu, kun näytöllä näkyy 100 %.
Akun voi ladata myös lisävarusteena hankittavalla
pöytälaturilla.

356 - FI PN MN232-C4 L | Käyttöopas

Tyypillinen akun latausaika täysin tyhjälle akulleon yhden akun kohdalla 3,5 tuntia, kaksoisakun
kohdalla 6,0 tuntia.
Huomautus: akku ei ehkä ota varausta vastaan,
jos se on täysin tyhjentynyt.

AKUN KÄYTTÖIKÄ

FreeStyle Comfort -akut on suunniteltu kestämään
vähintään 300 lataus- ja purkausjaksoa.

FreeStyle Comfort
Vaihto- ja tasavirtalähde
FreeStyle Comfort toimitetaan vakiona vaihtoja tasavirtalähteillä varustettuna.
Vaihtovirtalähde sisältää seuraavat osat:	a. Vaihtovirtayksikkö 100–240 V, 50–60 Hz.
	b. Virtajohto, joka kytketään
vaihtovirtayksikköön ja vastaavaan
liittimeen.

Tasavirtalähde muodostuu johdosta, joka kytketään
suoraan laitteeseen ja tasavirtaliitäntään.

2. Liitä vaihtovirtajohto vaihtovirtayksikköön.

VAIHTOVIRTA

Vaihtovirtaa käytetään kotona ja paikoissa, joissa
sitä on saatavana. FreeStyle Comfort toimii
vaihtovirralla. FreeStyle Comfort sisältää myös
vaihtovirta-akun, joka voidaan ladata aina, kun
vaihtovirtaa on saatavilla (paitsi lentokoneessa).
Akku latautuu riippumatta siitä, onko FreeStyle
Comfort päällä vai ei.
VAROITUS: ÄLÄ KÄYNNISTÄ
HAPPIRIKASTINTA ILMAN SIIHENASENNETTUA AKKUA.
FreeStyle Comfort -rikastimen kytkeminen
vaihtovirtaan:
1. Kytke vaihtovirtayksikkö FreeStyle Comfort-laitteen etupuolella oikeassa alakulmassa
olevaan liittimeen. Paina liitin kunnolla pohjaan,
jotta liitäntä on tiukka.

3. Työnnä vaihtovirtajohdon pistoke pistorasiaan.

PN MN232-C4 L | Käyttöopas FI - 357

FreeStyle Comfort
4. Käynnistä FreeStyle Comfort painamalla
virtapainiketta ja valitse haluamasi virtausasetus
lisäämällä virtausta (+) tai vähentämällä
virtausta (-).

4. Käynnistä FreeStyle Comfort painamalla
virtapainiketta ja valitse haluamasi virtausasetus
lisäämällä virtausta (+) tai vähentämällä
virtausta (-).
Huomautus: jos akku on ladattava uudelleen,
näytön akkukuvakkeessa näky yksi oranssi palkki.

TASAVIRTA

Tasavirtaa on saatavana autoissa, veneissä jamuissa moottoriajoneuvoissa. FreeStyle Comfort
toimii myös tasavirralla. FreeStyle Comfort lataa
akkupakkauksen tarvittaessa aina, kun tasavirtaa
on saatavilla. Akku latautuu riippumatta siitä, onkohappirikastin käynnistetty vai ei.
VAROITUS: ÄLÄ LIITÄ ULKOISTAVIRTALÄHDETTÄ ILMAN, ETTÄ
HAPPIRIKASTIMEN AKKU ON
ASENNETTUNA.
1. Käynnistä moottoriajoneuvo.
2. Työnnä tasavirtajohto laitteen etupuolella
oikeassa alareunassa olevaan liittimeen. Paina
liitin kunnolla pohjaan, jotta liitäntä on tiukka.

TÄRKEÄ HUOMAUTUS: Mahdollisen
tulipalon tai ylikuumenemisen estämiseksi
varmista, että 12 V:n tasavirtaliitäntä
on puhdas ja että tasavirtalähteen
sovitinpistoke sopii liitäntään kunnolla
ja pysyy tukevasti paikallaan.
TÄRKEÄ HUOMAUTUS: Tasavirtalähteen
metallipinnat voivat kuumentua ERITTÄINKUUMIKSI käytön aikana suuren virran vuoksi.
Anna pintojen jäähtyä riittävästi ennen kuin
kosketat niitä.
TÄRKEÄ HUOMAUTUS: Älä käytä tasavirtalähdettä
jatkojohtojen, jakajien tai sovittimien kanssa.
Muiden kuin hyväksyttyjen lisävarusteiden
käyttö voi johtaa niiden ylikuumenemiseen tai
vaurioitumiseen.
TÄRKEÄ HUOMAUTUS: Laitteen liittäminen 12 V:ntasavirtaliitäntään voi tyhjentää ajoneuvon akun,
jos moottori ei ole käynnissä, vaikka laitteen virta
olisi katkaistu.
TÄRKEÄ HUOMAUTUS: Älä käynnistä ajoneuvoa,
kun tasavirtalähde on liitettynä.

3. Työnnä tasavirtajohdon toinen pää
auton tasavirtapistokkeeseen tai muuhun
tasavirtapistokkeeseen.

358 - FI PN MN232-C4 L | Käyttöopas

FreeStyle Comfort
Puhdistus, ylläpito ja huoltoMÄÄRÄAIKAISKUNNOSSAPITO

Ennaltaehkäisevä huolto
Laitetta ei tarvitse huoltaa. Laitteen sisällä olevat
suodattimet on tarkoitettu kestämään laitteen
koko käyttöiän. Jos laite ei toimi kunnolla, laite
on palautettava palveluntarjoajalle korjattavaksi.
FreeStyle Comfort -happirikastimen sisäosien
huolto on annettava laitetoimittajan tehtäväksi.
Laiteen käyttäjä on vastuussa seuraavasta
huoltotoimenpiteestä.
• Ilmanottosuodattimet täytyy puhdistaa kerranviikossa.
VAROITUS: ÄLÄ YRITÄ AVATA TAIIRROTTAA HAPPIRIKASTIMENMUOVISUOJIA. LAITTEEN SAA AVATAVAIN PÄTEVÄ HUOLTOTEKNIKKO.

ILMANOTTOSUODATTIMETFreeStyle Comfort -rikastimeen otetaan
ilmaa laitteen edessä alareunassa sijaitsevien
ilmanottosuodattimien kautta. Suodattimet
estävät ilmassa olevan pölyn ja muiden suurien
partikkeleiden pääsyn laitteeseen.
Ilmanottosuodattimet täytyy puhdistaa vähintään
kerran viikossa. Laitetoimittaja saattaa neuvoa
puhdistamaan laitteen useammin käyttöolosuhteista
riippuen. Puhdista ilmanottosuodattimet toimimalla
seuraavasti.
1. Irrota kaksi suodatinta FreeStyle Comfort-happirikastimen edestä työntämällä ylös
kyseistä kielekettä ja vetämällä ne ulos.

VAROITUS: ÄLÄ KÄYTÄ HAPPIRIKASTINTA, JOS TULOILMASUODATIN EI
OLE PAIKALLAAN.

AKUSTA HUOLEHTIMINEN
Älä anna akun altistua nesteille.

Älä yritä purkaa tai avata akkua.
Älä altista akkua avotulelle.
Älä pudota akkua.
Pidä akku poissa lasten ulottuvilta.
Jos akkupakkaus on vahingoittunut, ota yhteyttä
laitetoimittajaan ennen käyttöä.
Älä aseta akkua mikroaaltouunin tai lieden lähelle.

AKUN SÄILYTTÄMINEN

Happirikastimen akun varauksen ei tarvitse
purkautua kokonaan ennen kuin akku ladataan
uudelleen. On suositeltavaa ladata happirikastimen
akku jokaisen käyttökerran jälkeen.
Älä altista akkua kuumuudelle, kuten
auringonpaisteelle. Älä jätä sitä autoon. Muutoin
akun suorituskyky voi heikentyä ja käyttöikä
lyhentyä. Lisäksi akku voi kuumentua, savuta,
syttyä tuleen tai räjähtää. Akkua tulee säilyttää
vakaassa ympäristössä, jossa suhteellinen
ilmankosteus alittaa 70 % ja jossa ei esiinnykorroosiota aiheuttavia kaasuja. Vältä esimerkiksiäkillisiä lämpötilan vaihteluita, koska ne saattavat
aiheuttaa kosteuden tiivistymistä ja ruostumista.
Jos akku on käyttämättä vähintään kuukauden,
on suositeltavaa ladata se 50–75-prosenttisesti(2–3 merkkivaloa palaa). Kun akku on ladattu,
irrota se happirikastimesta.

2. Pese molemmat suodattimet lämpimällä
saippualiuoksella ja vedellä.
3. Huuhtele suodattimet huolellisesti ja kuivaa
ylimääräinen vesi pehmeällä, nukkaamattomalla
kankaalla. Varmista, että suodattimet ovat täysin
kuivia, ennen kuin laitat ne takaisin.
4. Asenna takaisin puhdas, kuiva ja suodatin.
PN MN232-C4 L | Käyttöopas FI - 359

FreeStyle Comfort
PUHDISTAMINEN

Katkaise laitteesta virta ja irrota se vaihto- taitasavirtalähteestä ennen puhdistamista. ÄLÄ
suihkuta nesteitä suoraan ulkokoteloon. Käytä
kosteaa (ei märkää) liinaa tai sientä. Suihkuta
liinaan tai sieneen mietoa kotikäyttöön tarkoitettua
puhdistusaineliuosta kotelon ja virtaliittimien
puhdistamista varten. Noudata puhdistusaineen
valmistajan ohjeita.
VAROITUS: KATKAISE LAITTEESTAVIRTA JA IRROTA PISTOKESÄHKÖPISTORASIASTA ENNEN KUIN
PUHDISTAT LAITTEEN. MUUTOINSEURAUKSENA ON SÄHKÖISKUNJA PALOVAMMAN VAARA. VAINLAITETOIMITTAJA TAI PÄTEVÄHUOLTOTEKNIKKO SAA POISTAALAITTEEN KANNET TAI HUOLTAALAITETTA.
VAROITUS: HUOLEHDI, ETTÄ HAPPIRIKASTIN
TAI AKKU EI PÄÄSE KASTUMAAN JA ETTÄLAITTEEN SISÄÄN EI PÄÄSE NESTEITÄ. MUUTOINSEURAUKSENA VOI OLLA TOIMINTAHÄIRIÖ TAI
LAITTEEN SAMMUMINEN, JA SÄHKÖISKUN TAIPALOVAMMOJEN RISKI VOI LISÄÄNTYÄ.
VAROITUS: ÄLÄ SUIHKUTA NESTETTÄ SUORAAN
LAITTEESEEN. ESIMERKIKSI SEURAAVATKEMIKAALIT OVAT HAITALLISIA: ALKOHOLI JA
ALKOHOLIPOHJAISET TUOTTEET, VÄKEVÖITYYNKLOORIIN POHJAUTUVAT TUOTTEET(ETYLEENIKLORIDI) JA ÖLJYPOHJAISETTUOTTEET. HAPPIRIKASTIMEN MUOVIOSIA EISAA PUHDISTAA TÄLLAISILLA AINEILLA, KOSKA
NE VAURIOITTAVAT LAITETTA.
VAROITUS: PUHDISTA RUNKO, OHJAUSPANEELI
JA VIRTAJOHTO VAIN MIEDOLLA KOTIKÄYTTÖÖNTARKOITETULLA PUHDISTUSAINEELLA. LEVITÄ
SITÄ KOSTEALLA LIINALLA TAI SIENELLÄ. PYYHIPINNAT KUIVIKSI. ÄLÄ PÄÄSTÄ HAPPIRIKASTIMEN SISÄÄN MITÄÄN NESTETTÄ. KIINNITÄ ERITYISTÄ HUOMIOTA KANYYLIN HAPPILIITTIMEENJA VARMISTA, ETTÄ SE ON PÖLYTÖN, KUIVA JA
PUHDAS.

360 - FI PN MN232-C4 L | Käyttöopas

VAROITUS: PUHDISTA AKKU KOSTEALLA(EI MÄRÄLLÄ) LIINALLA TAI SIENELLÄ.
SUIHKUTA LIINAAN TAI PESUSIENEENMIETOA KOTIKÄYTTÖÖN TARKOITETTUAPUHDISTUSAINETTA. PUHDISTA AKKUKOTELO.
VAROITUS: ÄLÄ PUHDISTA AUTOMAATTISISSAPESUJÄRJESTELMISSÄ.

KANYYLIN VAIHTAMINEN

Noudata aina kanyylin valmistajan käyttöohjeita.
Vaihda nenäkanyyli ja happiletkut 3–6 kuukauden
välein (jos laite on päivittäisessä käytössä,3 kuukauden välein) kanyylin valmistajansuositusten mukaisesti sekä happirikastimen
toimittajan suositusten mukaisesti. Lääkärisi
tai happirikastimen toimittaja antaa puhdistusja vaihto-ohjeet.
Varaosia on saatavana happirikastimen toimittajalta.

FreeStyle Comfort
Hälytystilat
Prioriteetti

Hälytyksen
kuvaus1

Järjestelmän
toimintahäiriö,
akun korkea
lämpötila

Näyttö

Äänimerkki

Lisätietoja/toimenpide-ehdotus

Yksipiippaus
20 sekunnin
väleinLaite antaa hälytyksen, jos akun lämpötila
lähestyy turvallisuussyistä tapahtuvan
sulkemisen kynnysarvoa.
Jos sulkemisen kynnysarvo saavutetaan,laite sammuu. Jos ulkoinen virtalähde on
liitettynä, hälytysnäyttö tulee näkyviin jaäänimerkki jatkuu kahden minuutin ajan.
Testaa laite niin, että akku on huoneenlämpöinen. Jos ongelma jatkuu, ota yhteyttä
laitetoimittajaan.

2

Järjestelmän
toimintahäiriö,
korkealämpötila

Yksipiippaus
20 sekunnin
väleinLaite antaa hälytyksen, jos sisäinen lämpötila lähestyy sulkemisen kynnysarvoa.
Jos sulkemisen kynnysarvo saavutetaan,laite sammuu, hälytysnäyttö tulee näkyviin
ja äänimerkki jatkuu kahden minuutin ajan.
Varmista, että laitteen ilma-aukoissa ei ole
tukoksia. Jos ongelma jatkuu, ota yhteyttä
laitetoimittajaan.

3

Järjestelmän
toimintahäiriö,
korkea paine(korkea järjestelmäpaine tai
kompressorin
vika)

Yksi
piippaus
20 sekunnin
väleinJos sulkemisen kynnysarvo saavutetaan,laite sammuu, hälytysnäyttö tulee näkyviin
ja äänimerkki jatkuu kahden minuutin ajan.

4

Järjestelmän
toimintahäiriö,
virtausnopeus(OCD-venttiilivika,
kaasureitin
tukkeutuminen)Yksipiippaus
30 sekunnin
väleinJos ongelma jatkuu, ota yhteyttä
laitetoimittajaan.

5

Akun vähäisen virran
aiheuttama
sammuttaminenYksipiippaus
30 sekunnin
väleinHälyttää, kunnes akun virta loppuu ja laite
sammuu.

Jos ongelma jatkuu, ota yhteyttä
laitetoimittajaan.

Lataa akku.

PN MN232-C4 L | Käyttöopas FI - 361

FreeStyle Comfort
Prioriteetti

Hälytyksen
kuvaus6

Hengitystä
ei havaittu

Näyttö

Äänimerkki

Lisätietoja

1 piippaus
10 minuutin välein

Hengitystä ei havaittu 15 sekuntiin.
Aktivoituu ensimmäisen hengenvedon jälkeen tai kun lämmitysnäyttö sammuu sen
mukaan, kumpi ehto täyttyy aikaisemmin.
Tarkista kanyyli vääntymien tai vuotojen varalta. Lisää herkkyysasetusta. Jos ongelma
jatkuu, ota yhteyttä laitetoimittajaan.

7

Lämmitys

Ei mitään

Laite käynnistyy automaattisella annostelulla, kunnes hengitys havaitaan tai
lämmitys loppuu, sen mukaan, kumpi ehto
täyttyy aikaisemmin.

8

Alhainenpitoisuus

1 piippaus
10 minuutin välein
enintään
30 minuutin ajan

Jos ongelma jatkuu, ota yhteyttä laitetoimittajaan.

9

Akku vähissä-varoitus

Yksipiippaus,
ei toistoa

Lataa akku.

10

Hengitystaajuus ylitettyYksipiippaus,
ei toistoa

Hengitystaajuus ylittää 40 bpm.

11

Järjestelmän
toimintahäiriö,
puhallinvika

Ei mitään

Jos ongelma jatkuu, ota yhteyttä
laitetoimittajaan.

12

Järjestelmän
toimintahäiriö,
alhainenpaine

Ei mitään

Jos ongelma jatkuu, ota yhteyttä
laitetoimittajaan.

13

Järjestelmän
toimintahäiriö,
alhainenlämpötila

Ei mitään

Anna laitteen lämmetä huoneenlämpöiseksi. Jos ongelma jatkuu, ota yhteyttä
laitetoimittajaan.

362 - FI PN MN232-C4 L | Käyttöopas

Jos annos käynnistyy virheellisesti,
pienennä herkkyysasetusta. Jos ongelma
jatkuu, ota yhteyttä laitetoimittajaan.

FreeStyle Comfort
Prioriteetti

Hälytyksen
kuvausInfo

Käynnistys

Näyttö

Äänimerkki

Lisätietoja

1 infopiippaus

Näkyy näytössä käynnistyksen yhteydessä,
kuulet yhden äänimerkin, joka vahvistaa,
että äänimerkki toimii.
Käynnistysnäyttö kertoo laiteversion
oikeassa alakulmassa.

Info

Normaalikäyttönäyttö
autoSATtilassa(tasavirtaa ei
ole kytketty)

Ei mitään

Info

Normaalikäyttönäyttö
ja autoSAT
pois käytöstä(tasavirtaa ei
ole kytketty)

Ei mitään

Info

Asetustamuutettu

1 infopiippaus

Info

Kytke
ulkoinenvirtalähde
laitteen ollessa poissa
päältä.

Ei mitään

Info

Kytke
ulkoinenvirtalähde
laitteenollessakäynnissä.

Ei mitään

Info

Kytke
ulkoinenvirtalähde
irti laitteen
ollessapoissa päältä.

1 infopiippaus

Näyttää CAIRE-näytön 2 sekunnin ajan
ja sen jälkeen akun ja kytkennän tilan
5 sekunnin ajan, ennen kuin näyttö
sammuu.

Näyttää CAIRE-näytön 2 sekunnin ajan,
minkä jälkeen näyttö sammuu.

PN MN232-C4 L | Käyttöopas FI - 363

FreeStyle Comfort
Prioriteetti

Hälytyksen
kuvausNäyttö

Info

Irrotaulkoinen virtalähde laitteen
ollessa käynnissä (akkulaitteessa).

1 infopiippaus

Info

Valmiustilassa - kun
ylös- taialas-painiketta painetaan,näytössä
näkyy akun
ja latauksen
tila.

Ei mitään

Info

Lentokonetila
akkuvirralla

1 pitkäpiippausInfo

Lentokonetila,
kun kytkettynä

1 pitkäpiippaus

364 - FI PN MN232-C4 L | Käyttöopas

Äänimerkki

Lisätietoja

Näyttö näkyy 5 sekuntia ja sammuu sitten.

FreeStyle Comfort
Valikon selausohjeet
Voit avata valikon valmius- tai käyttötilassa.
Jos käyttötilassa ilmenee hälytystila, valikko
sulkeutuu automaattisesti. Jos mitään painiketta
ei paineta 20 sekuntiin, valikko sulkeutuu
automaattisesti. Valmiustilassa valikkopainiketta
voi olla tarpeen painaa kaksi kertaa, jotta laite
siirtyy käyttötilaan ja voit avata valikon.
Valikkonäyttöjä on neljä:
1. Tilanäyttö: sisältää laiteohjelmistonversionumeron, käyttöajan ja akun purkausjaksot

Lentokonetila

2. Lentokonetila (ON/OFF)
3. Herkkyys: säädettävä (1–5)
4. autoSAT (ON/OFF)

Herkkyys

Esimerkki tilanäytöstä

ASETUSVALIKON KÄYTTÖ
autoSAT

Voit avata asetusvalikon painamalla valikko-/
valintapainiketta
valmiustilassa.

, kun laite on käyttö- tai

Voit selata valikkoja painamalla painiketta.

Tilasivu

PN MN232-C4 L | Käyttöopas FI - 365

FreeStyle Comfort
TILASIVU

Jos haluat peruttaa lentokonetilan ottamisen

FreeStyle Comfortin tilasivulla on laiteohjelmistonversio, käyttötunnit ja liitetyn akun latausjaksojen
määrä.

käyttöön, paina-painiketta Cancel (Peruuta)-kuvakkeen ollessa näkyvissä.

Lentokonetila käytössä -asetus vahvistettu
Tilasivu

Käyttötunnit näytetään kokonaisina tunteina.
Akun purkausjaksojen määrä tarkoittaa sitä,
montako kertaa liitetty akku on purkautunut
ja ladattu uudelleen.

LENTOKONETILA
Lentokonetila on pois käytöstä, kun FreeStyle
Comfort toimitetaan.

Kun haluat poistaa lentotilan käytöstä, toista edelläolevat vaiheet ja vaihda asetukseksi OFF.
Kun näytössä näkyy Next (Seuraava) -kuvake
ja painat-painiketta, valikkosivu siirtyy
seuraavaan näyttövaihtoehtoon.
Lentotila on tarkoitettu käytettäväksi käyttötilassa.
Lentotila kytkeytyy automaattisesti pois käytöstä,jos laite jätetään valmiustilaan akkuvirralla.
Huomautus: lisätietoja lentotilasta on kohdassa
Lentotilan aktivointi.

HERKKYYSTILA

Lentokonetilan päänäyttö

Kytke tila päälle painamalla-painiketta
Change (Muuta) -kuvakkeen ollessa näkyvissä.
Arvoksi muuttuu sininen ON (Käytössä) -sana.

Kun FreeStyle Comfort toimitetaan, senoletusarvoisena herkkyysasetuksena on 2.
Tätä asetusta ei ole tarpeen muuttaa normaaleissa
olosuhteissa. Herkkyysasetukseksi voidaan valitaarvo välillä 1–5, jossa 1 on vähiten herkkä ja
5 herkin.

Herkkyystilanäyttö, jossa asetuksena 2
Lentokonetila käytössä -asetus valittu

Vahvista lentokonetilan kytkeminen käyttöön
painamalla-painiketta Save (Tallenna)-kuvakkeen ollessa näkyvissä. Arvoksi muuttuu
musta ON (Käytössä) -sana.

366 - FI PN MN232-C4 L | Käyttöopas

FreeStyle Comfort
Voit säätää herkkyystilaa painamalla-painiketta
Change (Muuta) -kuvakkeen ollessa näkyvissä.
Arvoksi muuttuu seuraava sininen numero.

Herkkyystilan asetus 3 valittu

Vahvista herkkyystilan arvoksi 3 painamalla-painiketta Save (Tallenna) -kuvakkeen ollessa
näkyvissä. Arvoksi muuttuu musta numero 3.
Voit peruuttaa herkkyystilan muutoksen painamalla

AUTOSAT-TILA
autoSAT-tila on käytössä, kun FreeStyle Comfort
toimitetaan.

autoSAT-tilan päänäyttö

Voit kytkeä tilan pois käytöstä painamalla-painiketta Change (Muuta) -kuvakkeen ollessa
näkyvissä. Arvoksi muuttuu sininen OFF(Pois käytöstä) -sana.

-painiketta Cancel (Peruuta) -kuvakkeen
ollessa näkyvissä.

autoSAT-tila poiss käytöstä -asetus valittu

Vahvista autoSAT-tilan kytkeminen pois käytöstä
Herkkyystilan asetus 3 vahvistettu

painamalla-painiketta Save (Tallenna)-kuvakkeen ollessa näkyvissä. Arvoksi muuttuu
musta OFF (Pois käytöstä) -sana.
Voit peruuttaa autoSAT-tilan ottamisen käyttöön-painiketta Cancel (Peruuta)
painamalla-kuvakkeen ollessa näytössä.

Herkkyystilan asetus 5 (Max) vahvistettu

Kun näytössä näkyy Next (Seuraava) -kuvake
ja painat-painiketta, valikkosivu siirtyy
seuraavaan näyttövaihtoehtoon.
autoSAT-tila pois käytöstä -asetus vahvistettu

Kun haluat kytkeä autoSAT-tilan käyttöön, toista
edellä olevat vaiheet ja vaihda asetukseksi ON.
Kun näytössä näkyy Next (Seuraava) -kuvake
ja painat-painiketta, valikkosivu siirtyy
seuraavaan näyttövaihtoehtoon.
PN MN232-C4 L | Käyttöopas FI - 367

FreeStyle Comfort
Vianmääritys
OngelmaTodennäköinen syy

Ratkaisu

FreeStyle Comfort näyttää
hälytyksen ja antaa
katkonaista äänimerkkiä.

Lue Hälytystilat-taulukko.

Lue Hälytystilat-taulukko.

Happirikastin ei käynnisty.

Akku on tyhjä.

Käynnistä laite AC- tai DC-liitännän kautta.

Ei kytketty sähkövirtaan.

Varmista ulkoisten liitäntöjen pitävyys.

Yleinen toimintahäiriö.

Suorita täydellinen kuittaus irrottamalla akku
ja mahdolliset ulkoiset AC- tai DC-virtalähteet.
Asenna akku sitten uudelleen. Kuittaus
on valmis. Liitä laite tarvittaessa ulkoiseen
virtalähteeseen.
Ota yhteys laitteen toimittajaan ja ota
tarvittaessa käyttöön vaihtoehtoinen happilähde.

Kaikki muut ongelmat

Suorita täydellinen kuittaus irrottamalla akku
ja mahdolliset ulkoiset AC- tai DC- virtalähteet.
Asenna akku sitten uudelleen. Kuittaus
on valmis. Liitä laite tarvittaessa ulkoiseen
virtalähteeseen.
Ota käyttöön vaihtoehtoinen happilähde ja ota
yhteys laitteen toimittajaan.

Huomautus: osa hälytyksistä aiheuttaa laitteen sammumisen.
Huomautus: sähkökatkoksen aikana hälytysjärjestelmä nollautuu (poistaa kaikki aiemmat hälytykset) ja näyttää
vain viimeisimmän hälytyksen, jos hälytys on edelleen voimassa, kun sähköä taas saa.

368 - FI PN MN232-C4 L | Käyttöopas

FreeStyle Comfort
Tietoja matkustamisesta
FreeStyle Comfort noudattaa Yhdysvaltainilmailuhallinto FAA:n ohjeita, jotka koskevat
laitteen käyttöä liikennelentokoneissa. Ennen
matkustamista esimerkiksi junassa, lentokoneessa
tai laivassa:• Varmista, että happirikastin on hyvässä
käyttökunnossa.
• Ota yhteyttä liikennöitsijään vähintään 48 tuntia
ennen lähtöä ja ilmoita, että matkustat FreeStyle
Comfort mukanasi.
On suositeltavaa ottaa lisäakkuja mukaan matkalle.
Jos matkustat lentokoneella, varmista, että
akussa riittää virtaa esimerkiksi lähtöselvitystä
ja turvatarkastusta varten. Useimmat lentoyhtiöt
edellyttävät, että sinulla on mukana riittävä määrä
ladattuja akkuja, jotta akkuvirta riittää laitteen
käyttämiseen vähintään 150 prosenttia odotetusta
enimmäismatkustusajasta (lentoajat, maassaoloaika,
aika ennen ja jälkeen lennon, yhteyslennot ja
odottamattomat viivästykset).
Huomautus: jos matkustat lentäen, on suositeltavaa,
että ylimääräiset akut ovat mukana turvallisessa
kantolaukussa.

1. Käynnistä happirikastin VAIN akkutilassa
niin, ettei laite ole kytkettynä ulkoiseen
virtalähteeseen.

2. Ota lentokonetila käyttöön jollakin seuraavista
menetelmistä:
a. Ota lentotila käyttöön noudattamalla ohjeita,
jotka annetaan kohdassa Valikon selausohjeet.

b. Pidä ylös- ja alas-painikkeita painettuna
5 sekuntia, kunnes kuuluu pitkä piippaus
ja näyttöön tulee lentokonekuvake.

FAA:N MÄÄRÄYKSET

Vara-akut tulee suojata erikseen oikosulkujen
estämiseksi pitämällä ne alkuperäisessä
tuotepakkauksessa tai muutoin eristämällä navat,
esimerkiksi teippaamalla jokaisen akun navat
tai sijoittamalla ne erilliseen muovipussiin tai
suojapussiin, ja kuljettamalla ne käsimatkatavaroissa.

LENTOTILAN AKTIVOINTI
Akkuvirta
Kannettavaa happirikastinta (POC) voi käyttää
lentokoneessa normaalisti mutta VAIN akkuvirralla,
kun ulkoista virtalähdettä ei ole kytketty
Ulkoinen virtalähde
Jos lentokoneessa on käytössä sähköpistorasia
ja sen sijainti soveltuu happirikastimen
virtalähteelle, sitä voidaan käyttää laitteen akkujen
varajärjestelmänä. Tällöin on ennen akkuvirrasta
lentokoneen sähkövirtaan siirtymistä toimittava
seuraavasti:

3. Kun lentokonekuvake on näytössä, kytke laite
sähkövirtaan.

Huomautus: lentokoneen rajoitusten vuoksi akkuja
ei saa ladata lentotilassa.

PN MN232-C4 L | Käyttöopas FI - 369

FreeStyle Comfort
Tarvikkeet

LAITTEEN ASETTAMINEN
KANTOLAUKKUUN

Käytä yhdessä happirikastimen kanssa vain
seuraavia CAIREn toimittamia tarvikkeita.
Muiden kuin seuraavassa lueteltujen tarvikkeiden
käyttäminen voi heikentää happirikastimen tehoa
ja/tai turvallisuutta.

1. Aseta laite laukun pohjalle. Kohdista suodattimet
ja akun aukot oikein.

FreeStyle Comfort -vakiovarusteet
Kantolaukku

MI396-1

Reppu

MI459-1Olkahihna

MI397-1

Vaihtovirtalähde(kaikki johdot mukana)

PW036-1S

Tasavirtajohto

CD041-1

Ulkoinen suurten hiukkasten
suodatin (oikea)

FI226-1

Ulkoinen suurten hiukkasten
suodatin (vasen)

FI227-1

2. Kiinnitä alaosan vetoketju.

Valinnaiset lisätarvikkeet
8-kennoinen lisäakkupakkaus

BT034-1S

16-kennoinen akkupakkaus

BT035-1S

Pöytälaturi

BT036-1

Kannettava tarvikelaukku

MI372-2

Huomautus: Muiden tarvikkeiden mukana voidaan
toimittaa tietyssä maassa käytettävä pistoke.
Voit tilata muita tarvikkeita ottamalla yhteyden
CAIREen tai happirikastimen toimittajaan.

Kantolaukun ohjeet
FreeStyle Comfort laitetaan kantolaukkuun
ja laukku suljetaan seuraavasti:

370 - FI PN MN232-C4 L | Käyttöopas

OLKAHIHNAN KIINNITTÄMINEN
1. Olkahihna voidaan kiinnittää joko suoraan
laitteeseen tai laitteen kantolaukkuun.
2. Kiinnitä hihna laitteen tai laukun renkaaseen.

FreeStyle Comfort
Hävittäminen ja uudelleenkäyttö
Palauta aina FreeStyle Comfort kaikkine osineenlaitetoimittajalle hävitettäväksi. Voit myös
ottaa yhteyttä paikalliseen kaupungin tai
kunnan viranomaiseen ja kysyä ohjeita akun
asianmukaisesta hävittämisestä. Laite voidaan
siirtää käyttäjältä toiselle, mutta se on palautettava
laitetoimittajalle uudelleenkäsittelyä varten ennen
käyttäjän vaihtamista.

WEEE ja RoHS
Tämä symboli muistuttaa laitteen omistajaa
siitä, että elinkaarensa lopussa laite on kierrätettävä sähkö- ja elektroniikkalaiteromudirektiivin(WEEE) mukaisesti.

Materiaalit, joiden kanssa käyttäjä
joutuu suoraan kosketukseen• Happirikastimen kotelo: PC/ABS• Happirikastimen ohjauspaneeli: EBG7-polyesteri
tai vastaava• Hapen syöttösuutin: Alumiini• Tyyppikilpi: Lexan• Johtoliitännät: Polykarbonaatti/vinyylikloridi• Virtajohto: PVC, metalli• Virtalähde: Lexan 940 (polykarbonaatti)• Akkupakkaus: PC/ABS• Akun ja virtalähteen tarrat: Polyesterikalvo• Happirikastimen kantolaukku: Cordura,
neopreeni, nylonvuori• Hihna: Polypropeeni, asetyyli, vinyyli, cordura,
neopreeni

Tuotteemme täyttävät vaarallisia aineita koskevan
RoHS-direktiivin vaatimukset. Tuotteissamme onvain vähäisiä jäämiä lyijystä ja muista vaarallisista
aineista.
TÄRKEÄ HUOMAUTUS: Hävitä
akku laitetoimittajan tai paikallisten
viranomaisten kierrätysohjeiden
mukaisesti.
TÄRKEÄ HUOMAUTUS: Hävitä molekyylisiivilä
aina kaikkien liittovaltion, osavaltion ja paikallisten
ympäristömääräysten mukaisesti.

PN MN232-C4 L | Käyttöopas FI - 371

FreeStyle Comfort
EMC-testaus
Ohjeistus ja valmistajan ilmoitus – sähkömagneettinen säteily
FreeStyle Comfort on tarkoitettu käytettäväksi alla määritetyssä sähkömagneettisessa ympäristössä.
FreeStyle Comfort -asiakkaan tai -käyttäjän on varmistettava, että sitä käytetään sellaisessa ympäristössä.
Säteilytesti
Radiotaajuussäteily
CISPR 11
Radiotaajuussäteily
CISPR 11
Harmonisoidut päästöt
IEC 61000-3-2
Jännitteenvaihtelut/kohinapäästöt
IEC 61000-3-3

Vastaavuus

Sähkömagneettinen ympäristö – ohjeistus

Ryhmä 1FreeStyle Comfort käyttää radiotaajuusenergiaa vain sisäisiin
toimintoihinsa. Tämän vuoksi radiotaajuussäteily on hyvin matalaa
eikä se todennäköisesti häiritse lähellä olevia elektronisia laitteita.

Luokka B
Luokka A
Vastaa372 - FI PN MN232-C4 L | Käyttöopas

FreeStyle Comfort soveltuu käytettäväksi kaikissa tiloissa,
kuten kotona ja tiloissa, jotka on liitetty suoraan julkiseen
pienjänniteverkkoon, joka syöttää sähköä kotitalouksilletarkoitettuihin rakennuksiin.

FreeStyle Comfort
Ohjeistus ja valmistajan ilmoitus – sähkömagneettinen häiriönsieto

FreeStyle Comfort on tarkoitettu käytettäväksi alla määritetyssä sähkömagneettisessa ympäristössä. FreeStyle
Comfort -asiakkaan tai -käyttäjän on varmistettava, että sitä käytetään sellaisessa ympäristössä.
ImmuniteettitesIEC 60601 -testaustaso Vastaavuustaso
Sähkömagneettinen ympäristö –tausohjeistus
Sähkömagneettinen
Lattian tulee olla puuta, betonia tai
ympäristö –±8 kV sähköliitäntä±8 kV sähköliitäntä
keraamilaattaa. Jos lattia on peitettyohjeistus±15 kV ilman purkaus±15 kV ilman purkaus synteettisellä materiaalilla, suhteellisen
kosteuden on oltava vähintään 30 %.
IEC 61000-4-2
Transienttipurskeen ±2 kV virtakaapelitsieto
IEC 61000-4-4

Syöksyaalto
IEC 61000-4-5±2 kV virtakaapelit±1 kV tulo-/lähtökaapelit Ei mitään±1 kV kaapelien välillä±2 kV yhteismuotoinen
jännite vaihtovirtajohdoissa±1 kV kaapelien välillä±1 kV eromuotoinen
jännite vaihtovirtajohdoissa±2 kV yhteismuotoinen
jännite ulkoisilla
I/O-linjoilla
0 % UT 0,5 jakson ajan 0 % UT 0,5 jakson ajan
(0°, 45°, 90°, 135°,(0°, 45°, 90°, 135°,
180°, 225°, 270°, 315°) 180°, 225°, 270°, 315°)

Jännitteen vajoamat, lyhyet
katkokset ja
jännitteenvaihteluttulosähkökaapeleissa, IEC 61000-4-11

0 % UT 1 jakson ajan
(0°)

0 % UT 1 jakson ajan
(0°)

70 % UT (30 %:n pudotus UT:ssä)T:ssä) 25/30
jakson ajan (0°)

70 % UT (30 %:n pudotus UT:ssä)T:ssä) 25/30
jakson ajan (0°)

Verkkovirran laadun tulee olla ammattija kotihoidon hoitolaitosympäristöille
tarkoitettua.

Verkkovirran laadun tulee olla ammattija kotihoidon hoitolaitosympäristöille
tarkoitettua.

Verkkovirran laadun tulee olla ammattija kotihoidon hoitolaitosympäristöille
tarkoitettua. Jos FreeStyle Comfort-käyttäjä edellyttää jatkuvaa toimintaa
virtakatkojenkin aikana, tällöin virta on
suositeltavaa syöttää FreeStyle Comfort-laitteeseen keskeytymättömästä
virtalähteestä (UPS) tai akusta.

0 % UT jakson ajan (0°) 0 % UT jakson ajan (0°)
Verkkotaajuinen
Verkkotaajuisten magneettikenttien(50/60 Hz)
3 A/m
3 A/m
tason pitäisi vastata koti- taimagneettikenttä
sairaalarakennuksen tyypillistä tasoa.
IEC 61000-4-8
HUOMAA UT tarkoittaa AC-verkkojännitettä ennen testaustason soveltamista.

HUOMAA Joissakin tapauksissa FreeStyle Comfort voi nollautua täydellisen sähköstaattisen purkauksen seurauksena, jolloin normaali toiminta on palautettava täydellisellä kuittauksella. Katso lisätietoja kohdasta Vianmääritys.

PN MN232-C4 L | Käyttöopas FI - 373

FreeStyle Comfort
Ohjeistus ja valmistajan ilmoitus – sähkömagneettinen häiriönsieto

FreeStyle Comfort on tarkoitettu käytettäväksi alla määritetyssä sähkömagneettisessa ympäristössä.
FreeStyle Comfort -asiakkaan tai -käyttäjän on varmistettava, että sitä käytetään sellaisessa ympäristössä.
Immuniteettites- IEC 60601 -testaustaso Vastaavuustaso
Sähkömagneettinen ympäristö –tausohjeistus
10 Vrms
Johdettu radio10 Vrms
Jos kannettavia ja siirrettäviä
taajuusradiotaajuuslaitteita käytetään FreeStyle
150 kHz – 80 MHz
Comfort -happirikastimen jonkin
IEC 61000-4-6
osan lähellä, mukaan lukien kaapelit,
etäisyyden on oltava vähintään kyseisen
lähettimen taajuuden perusteella laskettu
10 V/m
10 V/msuositeltu etäisyys.
Säteilevä
80–2 700 MHz
radiotaajuus
IEC 61000-4-3

Suositeltu etäisyysd= 1.2 P= 1.2 P = 80–800 MHzd= 1.2 P = 800–2 700 MHzd
jossa P on lähettimen suurin lähtöteho
watteina (W) lähettimen valmistajan
ilmoituksen mukaisesti ja d on suositeltu
etäisyys metreinä (m).
Kiinteiden radiotaajuuslähettimien
kentänvoimakkuuksien (paikan
sähkömagneettisen mittauksen a
mukaisesti) on oltava alhaisempia
kuin kunkin taajuusalueen
säädöstenmukaisen tason. b
Häiriöitä voi esiintyä, jos lähellä on
seuraavalla symbolilla merkittyjä laitteita:

HUOMAA 1 Korkea taajuusalue pätee tehoilla 80 MHz ja 800 MHz.
HUOMAA 2 Tämä ohjeistus ei ehkä päde kaikissa tilanteissa. Sähkömagneettiseen johtumiseen vaikuttavat
pintojen, esineiden ja ihmisten aiheuttama imeytyminen ja heijastuminen.
aKiinteiden lähettimien, kuten matka- ja langattomien puhelinten sekä satelliittiradioiden asemien, amatööriradioiden,
AM- ja FM-radiolähetysten ja televisiolähetysten kentänvoimakkuuksia ei voi teoreettisesti ennakoida tarkasti.
Kiinteän radiotaajuuslähettimen aiheuttaman sähkömagneettisen ympäristön arvioimiseksi on suoritettava paikan
sähkömagneettinen mittaus. Jos FreeStyle Comfort -happirikastimen käyttöympäristön kenttävoimakkuus ylittää
yllä esitetyn sovellettavan radiotaajuuden vaatimustenmukaisuustason, FreeStyle Comfort -happirikastinta tulee
tarkkailla sen normaalin toiminnan varmistamiseksi. Jos toiminnassa havaitaan poikkeavuuksia, lisätoimenpiteet,kuten FreeStyle Comfort -laitteen uudelleensuuntaus tai -sijoitus, voivat olla tarpeen.
bTaajuusalueella 150 kHz – 80 MHz kentänvoimakkuuksien on oltava alle 10 V/m.

374 - FI PN MN232-C4 L | Käyttöopas

FreeStyle Comfort
Suositellut etäisyydet kannettavien ja siirrettävien radiotaajuusviestintälaitteiden sekä
FreeStyle Comfort -happirikastimen välillä
FreeStyle Comfort on tarkoitettu käytettäväksi sähkömagneettisessa ympäristössä, jossa radiotaajuuksien säteilyhäiriöitä kontrolloidaan. FreeStyle Comfort -asiakas tai käyttäjä voi auttaa estämään sähkömagneettisia häiriöitä
säilyttämällä alla olevan suosituksen mukaisen vähimmäisvälimatkan kannettavan ja siirrettävän radiotaajuuslaitteen (lähettimet) ja FreeStyle Comfort -happirikastimen välillä viestintälaitteen enimmäistehon mukaan.
Lähettimen suurin
Etäisyys lähettimen taajuuden mukaisesti
nimellislähtöteho
m
150 kHz – 80 MHz

80–800 MHz

800 MHz – 2,5 GHzW= 1.2 P= 1.2 P= 1.2 P
d
d
d
0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
Jos lähettimen suurinta lähtötehoa ei ole mainittu alla, suositeltu etäisyys d metreinä (m) voidaan arvioida yhtälöllä
lähettimen taajuuden perusteella, kun P on lähettimen suurin lähtöteho watteina (W) lähettimen valmistajan
ilmoituksen mukaisesti.
HUOMAA 1 Korkean taajuusalueen etäisyys pätee tehoilla 80 MHz ja 800 MHz.
HUOMAA 2 Tämä ohjeistus ei ehkä päde kaikissa tilanteissa. Sähkömagneettiseen johtumiseen vaikuttavat
pintojen, esineiden ja ihmisten aiheuttama imeytyminen ja heijastuminen.

Luokitus

Suojaus sähköiskuilta (tyyppi):
Luokka II	Laite on suojattu sähköiskuja			vastaan kaksoiseristämällä.
Suojausaste sähköiskua vastaan:	Tyyppi BF Laitteen suojaustaso sähköiskua
vastaan seuraavin osin:
		1) sallittu vuotovirta		2) maadoituksen luotettavuus
		(jos sellainen on).
		Laitetta ei ole tarkoitettu käytettäväksi suoraan
sydämeen.
Riippumaton testaaminen lääkinnällisiä laitteitakoskevan standardin mukaisesti:
	Täyttää testien mukaan• IEC 60601-1 versio 3.1: Sähkökäyttöiset
lääkintälaitteet - osa 1, Yleiset
turvallisuusvaatimukset.
• CAN/CSA C22.2 nro 60601-1-8 versio 2.1,
Sähkökäyttöiset lääkintälaitteet - osa 1:
Yleiset turvallisuusvaatimukset
Suojattu mahdollisia sähkömagneettisia ja muita laitteen sekä muiden laitteiden välisiä häiriöitä vastaan.
	• Täyttää testien mukaan standardin
EN 60601-1-2 /IEC 60601-1-2:
Versio 4 vaatimukset.

	• Täyttää testien mukaan standardin RTCA/
DO160 osa 21 luokka M vaatimukset.
• CISPR 11 / EN 55011, luokka B, ryhmä 1,
	"Industrial, Scientific, and Medical (ISM)	Equipment"• FCC osa 15, kohta B – luokka B	Tahattomat jäähdyttimet
IP22 – Suojaus sormien tai muiden vastaavien
esineiden läpimitaltaan suurempien kuin
12,5 mm sisäänpääsyltä. Pystysuoraan
tippuva vesi ei ole haitallista, vaikka kotelo on
kallistuneena 15 astetta normaalista asennostaan.
Sallittu puhdistus- ja infektioidenestomenetelmä:
Lisätietoja on FreeStyle-käyttöohjeen
puhdistamisesta, hoidosta ja kunnossapidosta
kertovassa osassa.
Turvallisuusaste herkästi syttyvien anestesiakaasujen
läheisyydessä:	Laite ei sovellu tällaiseen käyttöön.
Toimintatila: jatkuva käyttö

PN MN232-C4 L | Käyttöopas FI - 375

FreeStyle ComfortΣτοιχεία ελέγχου χρήστη και ενδείξεις κατάστασης συστήματος
ISO 7000Εγχειρίδιο χειριστή, οδηγίες
λειτουργίας. Reg. # 1641

ISO 7010: Γραφικά σύμβολα — Χρώματα ασφάλειας
και σήματα ασφάλειας — Καταχωρημένα σήματα
ασφάλειαςΗ χώρα και η ημερομηνία κατασκευής.
Το «CC» προσδιορίζει τον κωδικό
χώρας δύο γραμμάτων της χώρας
κατασκευής. Η ημερομηνία κατασκευής
είναι σε μορφή ΕΕΕΕ-MM-ΗΗ.
Reg. # 6049

Απαιτείται να διαβαστεί το εγχειρίδιο
οδηγιών. Reg. # M002
Μακριά από γυμνές φλόγες, φωτιά,σπίθες. Απαγορεύονται οι ανοιχτές
πηγές ανάφλεξης και το κάπνισμα.
Reg. # P003
Μην καπνίζετε κοντά στη μονάδαή κατά τη διάρκεια της λειτουργίας τηςμονάδας. Reg. # P002Φυλάσσετε μακριά από τη βροχή,διατηρείτε στεγνό. Reg. # 0626
Όριο στοίβαξης ανά αριθμό, όπου το«n» δηλώνει τον μέγιστο επιτρεπόμενο
αριθμό μονάδων. Reg. # 2403
Αριθμός καταλόγου. Reg. # 2493

Προειδοποίηση. Reg. # W001
IEC 60417
Εφαρμοζόμενο μέρος τύπου BF (βαθμός
προστασίας από ηλεκτροπληξία). Reg.
# 5333Σειριακός αριθμός. Reg. # 2498
Εύρος θερμοκρασίας αποθήκευσης
ή λειτουργίας. Reg. # 0632
Εύρος υγρασίας αποθήκευσης.
Reg. # 2620
Περιορισμός ατμοσφαιρικής πίεσης.
Reg. # 2621
Επωνυμία και διεύθυνση κατασκευαστή.
Reg. # 3082
Με αυτή την όψη προς τα πάνω.
Reg. # 0623

Εξοπλισμός κλάσης II, διπλά μονωμένος
Reg. # 5172
Κανονισμός ΕΕ 2017/745
Εξουσιοδοτημένος αντιπρόσωπος στηνΕυρωπαϊκή ΚοινότηταΕάν η ετικέτα αποκλειστικού
αναγνωριστικού τεχνολογικού
προϊόντος (UDI) φέρει το σύμβολοCE####, το τεχνολογικό προϊόν
συμμορφώνεται με τις απαιτήσεις τωνσχετικών ιατροτεχνολογικών προϊόντων.
Το σύμβολο CE#### υποδεικνύει τοναριθμό κοινοποιημένου οργανισμού.
Επιπρόσθετα σύμβολαΜακριά από εύφλεκτα υλικά, λάδια καιγράσο.

Εύθραυστο, χειριστείτε με προσοχή.
Reg. # 0621

Μην αποσυναρμολογείτε.

Περιέχει επικίνδυνες ουσίες.
Reg. # 3723

Αύξηση της ροής.

Εισαγωγέας. Reg. # 3725

Μείωση της ροής.

Έξοδος O2

Κουμπί μενού.
Διακόπτης λειτουργίας. Reg. # 5009Συμμορφώνεται με τηνΚατηγορίαΜ του προτύπου RTCA/
DO-160Ενότητα 21.

376 - GR PN MN232-C4 L | Εγχειρίδιο χρήστηΟ κατασκευαστής του παρόντος POC
έχει καθορίσει ότι αυτό το τεχνολογικό
προϊόν συμμορφώνεται με όλα ταισχύοντα κριτήρια αποδοχής της FAA[Υπηρεσία Πολιτικής Αεροπορίας τωνΗΠΑ] και χρήσης επί αεροσκαφών.

FreeStyle ComfortΟ κατασκευαστής του παρόντος POC
έχει καθορίσει ότι αυτό το τεχνολογικό
προϊόν συμμορφώνεται με όλα ταισχύοντα κριτήρια αποδοχής της FAA καιτη χρήση επί αεροσκαφών.

Οδηγία του Συμβουλίου 2012/19/ΕΕΑΗΗΕΑυτό το σύμβολο υπενθυμίζει στους
κατόχους του εξοπλισμού να τονεπιστρέψουν σε εγκαταστάσεις
ανακύκλωσης στο τέλος της ωφέλιμηςζωής του, σύμφωνα με την Οδηγία για ταΑπόβλητα Ηλεκτρικού και Ηλεκτρονικού
Εξοπλισμού (ΑΗΗΕ). Τα προϊόντα μας
συμμορφώνονται με την οδηγία για τονπεριορισμό των επικίνδυνων ουσιών
(RoHS). Δεν περιέχουν περισσότεροαπό ιχνοποσότητες μολύβδου ή άλλων
επικίνδυνων υλικών.

Σύμβολο ανακύκλωσης
Εξουσιοδοτημένος αντιπρόσωπος στην
Ελβετία (ADD)Εάν η συσκευή φέρει το σήμα
UKCA του Ηνωμένου Βασιλείου,όπως φαίνεται με τον αριθμό
του κοινοποιημένου οργανισμού
UKCA####, αυτό το τεχνολογικό
προϊόν συμμορφώνεται με τουςκανονισμούς της UKCA.
21 CFR [Κώδικας Ομοσπονδιακών Κανονισμών]801.15:Η ομοσπονδιακή νομοθεσία τωνΗΠΑ επιτρέπει την πώληση αυτής τηςσυσκευής μόνο από ιατρό ή κατόπιν
εντολής ιατρού.

IEC 60601-1-8Τεχνικός συναγερμός χαμηλής
προτεραιότητας
ISO 15223-1Ιατροτεχνολογικό προϊόν. Reg. # 5.7.7Πιστοποίηση ETLΠιστοποιημένο για τις αγορές τωνΗΠΑ και του Καναδά, σύμφωνα με ταισχύοντα πρότυπα των ΗΠΑ και τουΚαναδά.
Περιέχει αναγνωριστικό της FCC [Ομοσπονδιακή
Επιτροπή Επικοινωνιών των ΗΠΑ]: WAP2001
Ανακοίνωση FCC
Αυτή η συσκευή μπορεί να περιέχει
CYBLE-022001-00, συμπεριλαμβανομένης τηςκεραίας 2450AT18B100 από την Johnson Technology,και συμμορφώνεται με την παρ. 15 των κανόνων
FCC. Η συσκευή πληροί τις απαιτήσεις για έγκριση
αρθρωτού πομπού, όπως περιγράφονται στη δημόσια
κοινοποίηση της FCC DA00-1407. Η λειτουργία τουπομπού υπόκειται στις ακόλουθες δύο προϋποθέσεις:(1) Η συσκευή δεν πρέπει να προκαλεί επιβλαβείς
παρεμβολές και (2) Αυτή η συσκευή πρέπει ναδέχεται οποιαδήποτε λαμβανόμενη παρεμβολή,συμπεριλαμβανομένης παρεμβολής που μπορεί ναπροκαλέσει ανεπιθύμητη λειτουργία. Το προϊόν μπορεί
να καλύπτεται από ένα ή περισσότερα διπλώματα
ευρεσιτεχνίας, των ΗΠΑ και διεθνή. Παρακαλούμε
επισκεφθείτε τον δικτυακό τόπο της εταιρείας μαςπαρακάτω για τον κατάλογο των εφαρμοστέων
διπλωμάτων ευρεσιτεχνίας. Διπλώματα ευρεσιτεχνίας:www.caireinc.com/corporate/patentsΑριθμός αποκλειστικού αναγνωριστικού
τεχνολογικού προϊόντος 5.7.10
A S/NZS 4417.1, AS/NZS 4417.2
Σήμα κανονιστικής συμμόρφωσης
ACMA - υποδεικνύει τη συμμόρφωση
με τους κανονισμούς ACMA[Αυστραλιανή Αρχή Επικοινωνιών καιΜέσων Μαζικής Ενημέρωσης] (ADD)Ασύρματη τεχνολογία
Τύποι σύνδεσης

SSP, iAP2, GATTΣυχνότητα2402 έως 2480 MHzΜέγιστη έξοδος ισχύος RF+4 dBmΕύρος λειτουργίας

10 m (Κατηγορία 2)Συνιστάται το μηχάνημα να βρίσκεται σε ελάχιστη απόσταση 1,1 cm (0,43 in.) από το σώμα κατά τη διάρκειατης λειτουργίας. Δεν ισχύει για τις μάσκες, τη σωλήνωση
αέρα ή τα παρελκόμενα.

Γραφικά σύμβολα για χρήση στον εξοπλισμό
IEC 60417
Συνεχές ρεύμα Reg. # 5031
Εναλλασσόμενο ρεύμα Reg. # 5032
IEC 60601-1
Με προστασία από κάθετη πτώση
σταγόνων
PN MN232-C4 L | Εγχειρίδιο χρήστη GR - 377

FreeStyle Comfort
Αποσυσκευασία του FreeStyle ComfortΕπαληθεύστε ότι όλα ταπαρατιθέμενα εξαρτήματα
περιλαμβάνονται στησυσκευασία. Εάν λείπουν
οποιαδήποτε εξαρτήματα,επικοινωνήστε με τον πάροχο
οξυγόνου σας αμέσως.

Τυπικά παρελκόμενα• Φορητός συμπυκνωτής οξυγόνου• Τσάντα μεταφοράς• Ιμάντας ώμου• Πακέτο μπαταρίας 8 στοιχείων• Τροφοδοτικό AC• Καλώδιο τροφοδοσίας AC• Τροφοδοτικό DC• Εγχειρίδιο χρήστη

Προαιρετικά παρελκόμενα• Πρόσθετο πακέτο μπαταρίας
8 στοιχείων• Συστοιχία μπαταριών
16 στοιχείων• Επιτραπέζιος φορτιστής• Τσάντα παρελκομένων
«carry-all»Γνωριμία με το FreeStyle ComfortΕξοικειωθείτε με τα βασικά χαρακτηριστικά του FreeStyle Comfort και της διεπαφής χρήστη.

ΠρόσοψηΠροσάρτηση ιμάντα

Διεπαφή χρήστη

Έξοδος
οξυγόνουΈξοδος αέραΕισαγωγή αέρα/φίλτραΕισαγωγή αέρα/φίλτραΓλωττίδα
υποδοχήςθύραςτροφοδοσίας
Πακέτο μπαταρίας
Κουμπί απελευθέρωσης
πακέτου μπαταρίας
378 - GR PN MN232-C4 L | Εγχειρίδιο χρήστη

FreeStyle ComfortΚάτοψη
Οθόνη ενδείξεων
Αύξηση της ροήςΚουμπί
λειτουργίας

Κουμπί μενού

Μείωση της ροήςΟθόνη ενδείξεων με μπαταρία
Ρύθμιση ροήςΈνδειξη μπαταρίας

Ένδειξη autoSATΕμφάνιση
συναγερμών

Ποσοστό
μπαταρίας πουαπομένειΈνδειξηχορήγησης
ενισχυμένης
δόσηςΟθόνη ενδείξεων με τροφοδοσία AC/DCΡύθμιση ροήςΠεδίο εμφάνισης
συναγερμών

Ένδειξη μπαταρίας

Ένδειξη φόρτισης

PN MN232-C4 L | Εγχειρίδιο χρήστη GR - 379

FreeStyle ComfortΤι είναι ο συμπυκνωτής οξυγόνουΟ αέρας που αναπνέουμε περιέχει περίπου 21%οξυγόνο, 78% άζωτο και 1% άλλα αέρια. ΣτοFreeStyle® Comfort®, ο αέρας του δωματίου
αναρροφάται μέσα στο μηχάνημα μέσω τωνανοιγμάτων εισαγωγής αέρα. Στη συνέχεια, περνά
μέσα από ένα απορροφητικό υλικό που ονομάζεται«μοριακό κόσκινο». Αυτό το υλικό ξεχωρίζει τοοξυγόνο από το άζωτο και επιτρέπει μόνο στο οξυγόνο
να περάσει. Το αποτέλεσμα είναι μια ροή οξυγόνου
υψηλής συγκέντρωσης που παρέχεται στον χρήστη.
Το FreeStyle Comfort συνδυάζει προηγμένες
τεχνολογίες συμπύκνωσης οξυγόνου και διατήρησης
οξυγόνου για τη δημιουργία μιας φορητής συσκευήςη οποία επιτρέπει μέγιστη φορητότητα καικινητοποίηση. Η προηγμένη χορήγηση παλμικής
ροής ανιχνεύει γρήγορα πότε εισπνέει ο χρήστης καιχορηγεί έναν παλμό οξυγόνου στην αρχή της κάθεεισπνοής. Η χορήγηση παλμικής ροής είναι ιδανική γιαδραστηριότητες και χρονικό διάστημα μακριά από τοσπίτι. Εξοικονομεί οξυγόνο και παρατείνει τη διάρκεια
ζωής της μπαταρίας, σε αντίθεση με τη συνεχή ροή.
Το FreeStyle Comfort είναι αρκετά ελαφρύ ώστε ναμεταφέρεται μέσω ενός ιμάντα ώμου. Μπορεί ναλειτουργεί με εναλλασσόμενο ρεύμα (AC) (επιτοίχια
παροχή), συνεχές ρεύμα (DC) (όχημα) ή ισχύ απόμπαταρία.

Προοριζόμενη χρήση
Ο συμπιεστής οξυγόνου CAIRE FreeStyle Comfortενδείκνυται για τη χορήγηση συμπληρωματικού
οξυγόνου. Η συσκευή δεν προορίζεται για ζωτική
υποστήριξη, ούτε παρέχει οποιεσδήποτε δυνατότητες
παρακολούθησης ασθενούς.

Γιατί ο γιατρός σαςσυνταγογράφησε οξυγόνο
Πολλοί άνθρωποι πάσχουν από μια ποικιλία
καρδιακών, πνευμονικών και άλλων αναπνευστικών
παθήσεων. Ένας σημαντικός αριθμός από αυτά ταάτομα μπορούν να ωφεληθούν από συμπληρωματική
οξυγονοθεραπεία στο σπίτι, όταν ταξιδεύουν ή ενώσυμμετέχουν σε καθημερινές δραστηριότητες μακριάαπό το σπίτι.
Το οξυγόνο είναι ένα αέριο που αποτελεί το 21%του αέρα του περιβάλλοντος που αναπνέουμε καιο οργανισμός μας εξαρτάται από μια σταθερή
παροχή για να λειτουργεί κανονικά. Ο γιατρός σαςσυνταγογράφησε μια συγκεκριμένη ροή ή ρύθμιση γιατην αντιμετώπιση της ιδιαίτερης αναπνευστικής σαςκατάστασης.

380 - GR PN MN232-C4 L | Εγχειρίδιο χρήστηΑν και το οξυγόνο είναι μια μη εθιστική θεραπεία,η αυθαίρετη οξυγονοθεραπεία μπορεί να είναι
επικίνδυνη. Πρέπει να ζητήσετε ιατρική συμβουλή
πριν χρησιμοποιήσετε αυτόν τον συμπυκνωτή
οξυγόνου. Ο πάροχος εξοπλισμού που παρέχει
τον εξοπλισμό οξυγόνου σας θα σας δείξει πώς ναρυθμίσετε τον συνταγογραφημένο ρυθμό ροής.

Προφίλ χειριστή
Οι συμπυκνωτές οξυγόνου προορίζονται για ναπαρέχουν συμπληρωματικό οξυγόνο σε χρήστες
που πάσχουν από δυσφορία λόγω παθήσεων πουεπηρεάζουν την αποτελεσματικότητα των πνευμόνων
τους να μεταφέρουν οξυγόνο από τον αέρα στηνκυκλοφορία του αίματος. Οι φορητοί συμπυκνωτές
οξυγόνου (POC) δεν αποθηκεύουν ή περιέχουν
οξυγόνο. Δεν χρειάζονται επαναπλήρωση και μπορούν
να επαναφορτίζονται όπου υπάρχει διαθέσιμη ισχύς
AC ή DC (εκτός εάν βρίσκονται σε αεροσκάφος).
Αυτό καθιστά τον χρήστη σχετικά αυτάρκη από τηνάποψη της χρήσης στο σπίτι, της κινητοποίησης (τόσοεντός όσο και εκτός του σπιτιού), της κινητικότητας
και του γενικού τρόπου ζωής. Η χρήση συμπυκνωτή
οξυγόνου απαιτεί συνταγή γιατρού.
Ο τυπικός χρήστης οξυγονοθεραπείας είναι ηλικίαςάνω των 65 ετών και πάσχει από μια ποικιλία
αναπνευστικών παθήσεων, συμπεριλαμβανομένηςτης χρόνιας αποφρακτικής πνευμονοπάθειας(ΧΑΠ). Οι χρήστες τυπικά έχουν καλές γνωστικές
ικανότητες και πρέπει να μπορούν να γνωστοποιούν
τυχόν δυσφορία. Εάν ο χρήστης δεν είναι σε θέσηνα γνωστοποιήσει τυχόν δυσφορία, ή δεν είναι σεθέση να διαβάσει και να κατανοήσει την επισήμανσηκαι τις οδηγίες χρήσης του συμπυκνωτή, τότε ηχρήση συνιστάται μόνο υπό την επίβλεψη κάπου
ατόμου που μπορεί να το πράξει. Εάν γίνει αισθητή
οποιαδήποτε δυσφορία κατά τη διάρκεια της χρήσης
του συμπυκνωτή, συνιστάται στους χρήστες ναεπικοινωνούν με τον πάροχο υγειονομικής περίθαλψής
τους. Συνιστάται επίσης στους χρήστες να έχουν
διαθέσιμο εφεδρικό οξυγόνο (δηλ. φιάλη οξυγόνου)σε περίπτωση διακοπής ρεύματος ή βλάβης τουσυμπυκνωτή. Δεν απαιτούνται άλλες μοναδικές
δεξιότητες ή ικανότητες χρήσης για τη χρήση τουσυμπυκνωτή.

FreeStyle ComfortΣημαντικό!
Οι οδηγίες ασφαλείας καθορίζονται ως εξής:ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΣΗΜΑΝΤΙΚΕΣ ΠΛΗΡΟΦΟΡΙΕΣ ΑΣΦΑΛΕΙΑΣ ΓΙΑ ΚΙΝΔΥΝΟΥΣ ΠΟΥΜΠΟΡΕΙ ΝΑ ΠΡΟΚΑΛΕΣΟΥΝ ΣΟΒΑΡΟ
ΤΡΑΥΜΑΤΙΣΜΟ.
ΠΡΟΣΟΧΗ: Σημαντικές πληροφορίες
για την πρόληψη ζημιάς στο FreeStyle
Comfort.
Σημείωση: Πληροφορίες που χρήζουν ιδιαίτερης
προσοχής.

Ενδείξεις χρήσης
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΕΊΝΑΙ ΠΟΛΎ ΣΗΜΑΝΤΙΚΌ ΝΑ ΕΠΙΛΈΓΕΤΕ ΜΌΝΟ ΤΟ ΣΥΝΤΑΓΟΓΡΑΦΗΜΈΝΟ ΕΠΊΠΕΔΟ ΟΞΥΓΌΝΟΥ. ΜΗΝΑΛΛΆΖΕΤΕ ΤΗΝ ΕΠΙΛΟΓΉ ΡΟΉΣ ΕΚΤΌΣΕΆΝ ΣΑΣ ΤΟ ΥΠΈΔΕΙΞΕ ΑΔΕΙΟΎΧΟΣ
ΚΛΙΝΙΚΌΣ ΓΙΑΤΡΌΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΤΟ ΠΑΡΟΝ ΤΕΧΝΟΛΟΓΙΚΟ ΠΡΟΪΟΝ
ΔΕΝ ΕΧΕΙ ΕΛΕΓΧΕΙ Ή ΕΓΚΡΙΘΕΙ ΓΙΑ ΧΡΗΣΗ ΜΕΠΑΙΔΙΑΤΡΙΚΟΥΣ ΑΣΘΕΝΕΙΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Η ΟΜΟΣΠΟΝΔΙΑΚΉ ΝΟΜΟΘΕΣΊΑ(ΗΠΑ) ΠΕΡΙΟΡΊΖΕΙ ΤΗΝ ΠΏΛΗΣΗ Ή ΜΊΣΘΩΣΗ ΑΥΤΉΣ
ΤΗΣ ΣΥΣΚΕΥΉΣ ΑΠΌ Ή ΚΑΤ' ΕΝΤΟΛΉ ΓΙΑΤΡΟΎ ΉΆΛΛΟΥ ΑΔΕΙΟΎΧΟΥ ΠΑΡΌΧΟΥ ΥΓΕΙΟΝΟΜΙΚΉΣ
ΠΕΡΊΘΑΛΨΗΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΑΥΤΉ Η ΜΟΝΆΔΑ ΔΕΝΠΡΟΟΡΊΖΕΤΑΙ ΝΑ ΧΡΗΣΙΜΟΠΟΙΕΊΤΑΙ ΓΙΑ ΖΩΤΙΚΉ
ΥΠΟΣΤΉΡΙΞΗ. ΕΠΙΠΡΟΣΘΕΤΗ ΠΑΡΑΚΟΛΟΥΘΗΣΗΜΠΟΡΕΙ ΝΑ ΑΠΑΙΤΗΘΕΙ ΓΙΑ ΟΠΟΙΟΝΔΗΠΟΤΕΧΡΗΣΤΗ, Ο ΟΠΟΙΟΣ ΔΕΝ ΕΙΝΑΙ ΣΕ ΘΕΣΗ ΝΑΕΠΙΚΟΙΝΩΝΗΣΕΙ ΤΗ ΔΥΣΦΟΡΙΑ ΚΑΤΑ ΤΗ ΧΡΗΣΗ ΤΟΥΠΑΡΟΝΤΟΣ ΤΕΧΝΟΛΟΓΙΚΟΥ ΠΡΟΪΟΝΤΟΣ. ΧΡΗΣΤΕΣΜΕ ΜΕΙΩΜΈΝΗ ΑΚΟΉ Ή/ΚΑΙ ΌΡΑΣΗ ΜΠΟΡΕΊ ΝΑΑΠΑΙΤΟΎΝ ΒΟΉΘΕΙΑ ΜΕ ΤΗΝ ΠΑΡΑΚΟΛΟΎΘΗΣΗ
ΤΩΝ ΣΥΝΑΓΕΡΜΏΝ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΟΙ ΡΥΘΜΊΣΕΙΣ ΛΕΙΤΟΥΡΓΊΑΣ
ΠΑΛΜΙΚΉΣ ΔΌΣΗΣ ΠΡΈΠΕΙ ΝΑ ΚΑΘΟΡΊΖΟΝΤΑΙ ΓΙΑΚΆΘΕ ΧΡΗΣΤΗ ΕΞΑΤΟΜΙΚΕΥΜΈΝΑ ΓΙΑ ΤΙΣ ΑΝΆΓΚΕΣ
ΤΟΥ ΣΕ ΑΝΆΠΑΥΣΗ, ΚΑΤΆ ΤΗ ΔΙΆΡΚΕΙΑ ΆΣΚΗΣΗΣΚΑΙ ΌΤΑΝ ΤΑΞΙΔΕΎΕΙ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΤΟ ΠΑΡΟΝ ΤΕΧΝΟΛΟΓΙΚΟ
ΠΡΟΪΟΝ ΠΡΟΟΡΙΖΕΤΑΙ ΓΙΑ ΧΡΗΣΗ ΣΤΗΝ ΚΑΤ'ΟΙΚΟΝ ΦΡΟΝΤΙΔΑ ΥΓΕΙΑΣ ΚΑΙ ΣΕ ΠΑΡΟΜΟΙΑΠΕΡΙΒΑΛΛΟΝΤΑ ΕΞΩΤΕΡΙΚΩΝ ΙΑΤΡΕΙΩΝ.

ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΟΡΙΣΜΕΝΕΣ ΑΝΑΠΝΕΥΣΤΙΚΕΣΠΡΟΣΠΑΘΕΙΕΣ ΤΟΥ ΑΣΘΕΝΟΥΣ ΜΠΟΡΕΙ ΝΑ ΜΗΝΕΝΕΡΓΟΠΟΙΗΣΟΥΝ ΤΗ ΛΕΙΤΟΥΡΓΙΑ ΔΙΑΤΗΡΗΣΗΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΤΟ FREESTYLE COMFORT ΔΕΝΠΡΟΟΡΙΖΕΤΑΙ ΓΙΑ ΧΡΗΣΗ ΜΕ ΤΡΑΧΕΙΟΤΟΜΗΜΕΝΟΑΣΘΕΝΗ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΕΓΚΥΕΣ Ή ΘΗΛΑΖΟΥΣΕΣΓΥΝΑΙΚΕΣ ΔΕΝ ΠΡΕΠΕΙ ΝΑ ΧΡΗΣΙΜΟΠΟΙΟΥΝΤΑ ΠΑΡΕΛΚΟΜΕΝΑ ΠΟΥ ΣΥΝΙΣΤΩΝΤΑΙ ΣΕ ΑΥΤΟΤΟ ΕΓΧΕΙΡΙΔΙΟ, ΚΑΘΩΣ ΜΠΟΡΕΙ ΝΑ ΠΕΡΙΕΧΟΥΝ
ΦΘΑΛΙΚΕΣ ΕΝΩΣΕΙΣ.

Αντενδείξεις χρήσης
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΣΕ ΟΡΙΣΜΕΝΕΣΠΕΡΙΠΤΩΣΕΙΣ, Η ΧΡΗΣΗ ΜΗΣΥΝΤΑΓΟΓΡΑΦΗΜΕΝΟΥ ΟΞΥΓΟΝΟΥΜΠΟΡΕΙ ΝΑ ΕΙΝΑΙ ΕΠΙΚΙΝΔΥΝΗ. ΑΥΤΗ ΗΣΥΣΚΕΥΗ ΠΡΕΠΕΙ ΝΑ ΧΡΗΣΙΜΟΠΟΙΕΙΤΑΙΜΟΝΟ ΟΤΑΝ ΕΧΕΙ ΣΥΝΤΑΓΟΓΡΑΦΗΘΕΙΑΠΟ ΓΙΑΤΡΟ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΤΟ ΠΑΡΟΝ ΤΕΧΝΟΛΟΓΙΚΟ
ΠΡΟΪΟΝ ΠΡΟΟΡΙΖΕΤΑΙ ΓΙΑ ΧΡΗΣΗ ΕΝΤΟΣ Ή ΠΛΗΣΙΟΝΠΕΡΙΒΑΛΛΟΝΤΩΝ ΜΑΓΝΗΤΙΚΟΥ ΣΥΝΤΟΝΙΣΜΟΥ,ΟΠΩΣ ΟΙ ΜΑΓΝΗΤΙΚΟΙ ΤΟΜΟΓΡΑΦΟΙ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΔΕΝ ΠΡΟΟΡΙΖΕΤΑΙ ΓΙΑ ΧΡΗΣΗ
ΥΠΟ ΤΗΝ ΠΑΡΟΥΣΙΑ ΕΥΦΛΕΚΤΩΝ ΑΝΑΙΣΘΗΤΙΚΩΝ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΌΠΩΣ ΚΑΙ ΜΕ ΚΆΘΕ ΗΛΕΚΤΡΙΚΆ
ΤΡΟΦΟΔΟΤΟΎΜΕΝΗ ΣΥΣΚΕΥΉ, Ο ΧΡΉΣΤΗΣ ΜΠΟΡΕΊΝΑ ΑΝΤΙΜΕΤΩΠΊΣΕΙ ΠΕΡΙΌΔΟΥΣ ΕΚΤΌΣ ΛΕΙΤΟΥΡΓΊΑΣΩΣ ΑΠΟΤΈΛΕΣΜΑ ΔΙΑΚΟΠΉΣ ΗΛΕΚΤΡΙΚΟΎ
ΡΕΎΜΑΤΟΣ, Ή ΤΗΣ ΑΝΆΓΚΗΣ ΝΑ ΑΝΑΘΈΣΕΙ ΤΟ ΣΈΡΒΙΣ
ΤΟΥ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ ΣΕ ΕΞΕΙΔΙΚΕΥΜΈΝΟ
ΤΕΧΝΙΚΌ. Ο ΣΥΜΠΥΚΝΩΤΉΣ ΟΞΥΓΌΝΟΥ ΔΕΝ ΕΊΝΑΙ
ΚΑΤΆΛΛΗΛΟΣ ΓΙΑ ΟΠΟΙΟΝΔΉΠΟΤΕ ΧΡΗΣΤΗ ΟΟΠΟΊΟΣ ΘΑ ΜΠΟΡΟΎΣΕ ΝΑ ΥΠΟΣΤΕΊ ΑΝΕΠΙΘΎΜΗΤΕΣΕΠΙΠΤΏΣΕΙΣ ΣΤΗΝ ΥΓΕΊΑ ΩΣ ΑΠΟΤΈΛΕΣΜΑ ΤΈΤΟΙΑΣΠΡΟΣΩΡΙΝΉΣ ΔΙΑΚΟΠΉΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΤΟ ΠΑΡΟΝ ΤΕΧΝΟΛΟΓΙΚΟ
ΠΡΟΪΟΝ ΔΕΝ ΕΧΕΙ ΕΓΚΡΙΘΕΙ ΓΙΑ ΧΡΗΣΗ ΣΕ ΟΛΑΤΑ ΠΕΡΙΒΑΛΛΟΝΤΑ. Η ΕΚΘΕΣΗ ΣΕ ΟΡΙΣΜΕΝΕΣΗΛΕΚΤΡΟΜΑΓΝΗΤΙΚΕΣ ΠΗΓΕΣ ΜΠΟΡΕΙ ΝΑ ΕΧΕΙ ΩΣΑΠΟΤΕΛΕΣΜΑ ΤΗΝ ΠΑΡΕΜΠΟΔΙΣΗ Ή ΤΗΝ ΑΚΟΥΣΙΑΑΠΟΔΟΣΗ ΤΟΥ ΤΕΧΝΟΛΟΓΙΚΟΥ ΠΡΟΪΟΝΤΟΣ.
ΣΤΗΝ ΠΕΡΙΠΤΩΣΗ ΠΑΡΕΜΠΟΔΙΣΗΣ, Η ΛΕΙΤΟΥΡΓΙΑ
ΤΟΥ ΤΕΧΝΟΛΟΓΙΚΟΥ ΠΡΟΪΟΝΤΟΣ ΘΑ ΠΡΕΠΕΙ ΝΑΔΙΑΚΟΠΕΙ ΑΜΕΣΩΣ.

ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΤΟ ΠΑΡΟΝ ΤΕΧΝΟΛΟΓΙΚΟ ΠΡΟΪΟΝ
ΠΡΟΟΡΙΖΕΤΑΙ ΓΙΑ ΧΡΗΣΗ ΑΠΟ ΤΟΝ ΑΣΘΕΝΗ ΠΟΥΛΑΜΒΑΝΕΙ ΟΞΥΓΟΝΟΘΕΡΑΠΕΙΑ.

PN MN232-C4 L | Εγχειρίδιο χρήστη GR - 381

FreeStyle ComfortΟδηγίες ασφαλείας
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΔΕΝ ΕΠΙΤΡΈΠΕΤΑΙ
ΚΑΜΊΑ ΤΡΟΠΟΠΟΊΗΣΗ ΑΥΤΟΎ ΤΟΥΕΞΟΠΛΙΣΜΟΎ
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΑΥΤΉ Η ΣΥΣΚΕΥΉ ΠΑΡΈΧΕΙ
ΟΞΥΓΌΝΟ ΥΨΗΛΉΣ ΣΥΓΚΈΝΤΡΩΣΗΣ, ΤΟ ΟΠΟΊΟ
ΠΡΟΆΓΕΙ ΤΗΝ ΤΑΧΕΊΑ ΚΑΎΣΗ. ΜΗΝ ΕΠΙΤΡΈΠΕΤΕΤΟ ΚΆΠΝΙΣΜΑ Ή ΤΙΣ ΑΝΟΙΧΤΈΣ ΦΛΌΓΕΣ ΣΤΟΝΊΔΙΟ ΧΏΡΟ ΜΕ (1) ΑΥΤΉΝ ΤΗ ΣΥΣΚΕΥΉ, Ή(2) ΟΠΟΙΟΔΉΠΟΤΕ ΠΑΡΕΛΚΌΜΕΝΟ ΠΟΥ ΦΈΡΕΙ
ΟΞΥΓΌΝΟ. ΑΠΟΤΥΧΊΑ ΤΉΡΗΣΗΣ ΑΥΤΉΣ ΤΗΣΠΡΟΕΙΔΟΠΟΊΗΣΗΣ ΜΠΟΡΕΊ ΝΑ ΟΔΗΓΉΣΕΙ ΣΕΠΥΡΚΑΓΙΆ, ΚΑΤΑΣΤΡΟΦΉ ΙΔΙΟΚΤΗΣΊΑΣ Ή/ΚΑΙ ΝΑΠΡΟΚΑΛΈΣΕΙ ΦΥΣΙΚΌ ΤΡΑΥΜΑΤΙΣΜΌ Ή ΘΆΝΑΤΟ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΛΕΙΤΟΥΡΓΕΊΤΕ ΤΗ ΜΟΝΆΔΑΣΕ ΠΕΡΙΟΡΙΣΜΈΝΟ Ή ΣΤΕΝΌ ΧΏΡΟ (ΔΗΛ. ΜΙΚΡΉ
ΤΣΆΝΤΑ Ή ΤΣΆΝΤΑ ΧΕΙΡΌΣ) ΌΠΟΥ Ο ΑΕΡΙΣΜΌΣ
ΜΠΟΡΕΊ ΝΑ ΕΊΝΑΙ ΠΕΡΙΟΡΙΣΜΈΝΟΣ. ΚΆΤΙ ΤΈΤΟΙΟ
ΜΠΟΡΕΊ ΝΑ ΠΡΟΚΑΛΈΣΕΙ ΥΠΕΡΘΈΡΜΑΝΣΗ ΤΟΥΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ ΚΑΙ ΝΑ ΕΠΗΡΕΆΣΕΙ
ΑΡΝΗΤΙΚΆ ΤΗΝ ΑΠΌΔΟΣΗ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Ο ΣΥΜΠΥΚΝΩΤΉΣ ΠΡΈΠΕΙ ΝΑΤΟΠΟΘΕΤΕΊΤΑΙ ΜΕ ΤΈΤΟΙΟΝ ΤΡΌΠΟ ΏΣΤΕ ΝΑΑΠΟΦΕΎΓΕΤΑΙ Ο ΚΑΠΝΌΣ, ΟΙ ΡΥΠΑΝΤΈΣ Ή ΟΙΑΝΑΘΥΜΙΆΣΕΙΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΣΕ ΠΕΡΊΠΤΩΣΗ ΠΟΥ ΣΥΜΒΕΊ
ΚΆΠΟΙΟ ΣΟΒΑΡΌ ΠΕΡΙΣΤΑΤΙΚΌ ΜΕ ΑΥΤΉΝ ΤΗΣΥΣΚΕΥΉ, Ο ΧΡΉΣΤΗΣ ΘΑ ΠΡΈΠΕΙ ΑΜΈΣΩΣ ΝΑΑΝΑΦΈΡΕΙ ΤΟ ΠΕΡΙΣΤΑΤΙΚΌ ΣΤΟΝ ΠΆΡΟΧΟ Ή/ΚΑΙΣΤΟΝ ΚΑΤΑΣΚΕΥΑΣΤΉ. ΩΣ ΣΟΒΑΡΌ ΠΕΡΙΣΤΑΤΙΚΌ
ΟΡΊΖΕΤΑΙ ΈΝΑΣ ΤΡΑΥΜΑΤΙΣΜΌΣ, ΘΆΝΑΤΟΣ ΉΗ ΠΙΘΑΝΌΤΗΤΑ ΠΡΌΚΛΗΣΗΣ ΤΡΑΥΜΑΤΙΣΜΟΎ/ΘΑΝΆΤΟΥ ΣΕ ΠΕΡΊΠΤΩΣΗ ΕΠΑΝΕΜΦΆΝΙΣΗΣ ΤΟΥΠΕΡΙΣΤΑΤΙΚΟΎ. Ο ΧΡΉΣΤΗΣ ΜΠΟΡΕΊ ΕΠΊΣΗΣ ΝΑΑΝΑΦΈΡΕΙ ΤΟ ΠΕΡΙΣΤΑΤΙΚΌ ΣΤΗΝ ΑΡΜΌΔΙΑ ΑΡΧΉΣΤΗ ΧΏΡΑ ΌΠΟΥ ΣΥΝΈΒΗ ΤΟ ΠΕΡΙΣΤΑΤΙΚΌ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Η ΧΡΉΣΗ ΟΡΙΣΜΈΝΩΝ
ΠΑΡΕΛΚΟΜΈΝΩΝ ΧΟΡΉΓΗΣΗΣ
ΟΞΥΓΌΝΟΥ ΠΟΥ ΔΕΝ ΚΑΘΟΡΊΖΟΝΤΑΙ ΓΙΑΑΥΤΌΝ ΤΟΝ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ
ΜΠΟΡΕΊ ΝΑ ΕΠΗΡΕΆΣΕΙ ΑΡΝΗΤΙΚΆ
ΤΗΝ ΑΠΌΔΟΣΗ. ΠΑΡΑΠΟΜΠΈΣ ΣΤΑΣΥΝΙΣΤΏΜΕΝΑ ΠΑΡΕΛΚΌΜΕΝΑ
ΠΑΡΈΧΟΝΤΑΙ ΣΕ ΑΥΤΌ ΤΟ ΕΓΧΕΙΡΊΔΙΟ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΕΆΝ Ο ΣΥΜΠΥΚΝΩΤΉΣ
ΟΞΥΓΌΝΟΥ ΈΧΕΙ ΠΈΣΕΙ, ΥΠΟΣΤΕΊ ΖΗΜΙΆ Ή ΕΚΤΕΘΕΊΣΕ ΝΕΡΌ, ΠΑΡΑΚΑΛΟΎΜΕ ΕΠΙΚΟΙΝΩΝΉΣΤΕ ΜΕΤΟΝ ΠΡΟΜΗΘΕΥΤΉ ΤΟΥ ΕΞΟΠΛΙΣΜΟΎ ΣΑΣ ΓΙΑΈΛΕΓΧΟ Ή ΠΙΘΑΝΉ ΕΠΙΣΚΕΥΉ ΤΟΥ ΤΕΧΝΟΛΟΓΙΚΟΎ
ΠΡΟΪΌΝΤΟΣ. ΜΗ ΧΡΗΣΙΜΟΠΟΙΕΊΤΕ ΤΟΝΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ ΕΆΝ ΈΧΕΙ ΥΠΟΣΤΕΊ ΖΗΜΙΆΤΟ ΚΑΛΏΔΙΟ Ή ΤΟ ΦΙΣ ΡΕΎΜΑΤΟΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΑΣΦΑΛΊΣΤΕ ΚΑΤΆΛΛΗΛΑ,ΠΡΟΣΔΈΣΤΕ ΜΕ ΙΜΆΝΤΑ Ή ΣΤΕΡΕΏΣΤΕ ΜΕ ΆΛΛΟΝ
ΤΡΌΠΟ ΤΟΝ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ ΌΤΑΝΒΡΊΣΚΕΣΤΕ ΣΕ ΈΝΑ ΌΧΗΜΑ ΚΑΤΆ ΤΗ ΔΙΆΡΚΕΙΑ
ΜΕΤΑΦΟΡΆΣ ΓΙΑ ΝΑ ΑΠΟΤΡΈΨΕΤΕ ΖΗΜΙΆ ΉΤΡΑΥΜΑΤΙΣΜΌ.
382 - GR PN MN232-C4 L | Εγχειρίδιο χρήστη

ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΟΙ ΚΑΝΟΝΙΣΜΟΊ ΤΟΥ ΥΠΟΥΡΓΕΊΟΥ
ΜΕΤΑΦΟΡΏΝ ΤΩΝ ΗΠΑ (DOT) ΚΑΙ ΤΩΝ ΗΝΩΜΈΝΩΝ
ΕΘΝΏΝ (UN) ΑΠΑΙΤΟΎΝ ΤΗΝ ΑΦΑΊΡΕΣΗ ΤΗΣΜΠΑΤΑΡΊΑΣ ΑΠΌ ΤΗ ΣΥΣΚΕΥΉ ΓΙΑ ΚΆΘΕ ΔΙΕΘΝΈΣ
ΑΕΡΟΠΟΡΙΚΌ ΤΑΞΊΔΙ ΌΤΑΝ Ο ΣΥΜΠΥΚΝΩΤΉΣ
ΟΞΥΓΌΝΟΥ ΕΛΈΓΧΕΤΑΙ ΩΣ ΑΠΟΣΚΕΥΉ. ΚΑΤΆ ΤΗΝΑΠΟΣΤΟΛΉ ΤΟΥ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ, ΗΜΠΑΤΑΡΊΑ ΠΡΈΠΕΙ ΕΠΊΣΗΣ ΝΑ ΑΦΑΙΡΕΊΤΑΙ ΑΠΌ ΤΗΣΥΣΚΕΥΉ ΚΑΙ ΝΑ ΣΥΣΚΕΥΆΖΕΤΑΙ ΚΑΤΆΛΛΗΛΑ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΛΕΙΤΟΥΡΓΕΙΤΕ ΑΥΤΟΝ ΤΟΝΕΞΟΠΛΙΣΜΟ ΧΩΡΙΣ ΝΑ ΕΧΕΤΕ ΠΡΩΤΑ ΔΙΑΒΑΣΕΙΚΑΙ ΚΑΤΑΝΟΗΣΕΙ ΤΟ ΠΑΡΟΝ ΕΓΧΕΙΡΙΔΙΟ. ΕΑΝ ΔΕΝΜΠΟΡΕΙΤΕ ΝΑ ΚΑΤΑΝΟΗΣΕΤΕ ΤΙΣ ΠΡΟΕΙΔΟΠΟΙΗΣΕΙΣΚΑΙ ΤΙΣ ΟΔΗΓΙΕΣ, ΕΠΙΚΟΙΝΩΝΗΣΤΕ ΜΕ ΤΟΝ ΠΑΡΟΧΟ
ΤΟΥ ΕΞΟΠΛΙΣΜΟΥ ΣΑΣ ΠΡΟΤΟΥ ΕΠΙΧΕΙΡΗΣΕΤΕ ΝΑΧΡΗΣΙΜΟΠΟΙΗΣΕΤΕ ΑΥΤΟΝ ΤΟΝ ΕΞΟΠΛΙΣΜΟ· ΣΕΑΝΤΙΘΕΤΗ ΠΕΡΙΠΤΩΣΗ, ΜΠΟΡΕΙ ΝΑ ΠΡΟΚΛΗΘΕΙΤΡΑΥΜΑΤΙΣΜΟΣ Ή ΥΛΙΚΗ ΖΗΜΙΑ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΕΆΝ ΑΙΣΘΆΝΕΣΤΕ ΔΥΣΦΟΡΊΑ ΉΠΑΡΟΥΣΙΆΖΕΤΕ ΕΠΕΊΓΟΥΣΑ ΙΑΤΡΙΚΉ ΚΑΤΆΣΤΑΣΗ,ΑΝΑΖΗΤΉΣΤΕ ΙΑΤΡΙΚΉ ΒΟΉΘΕΙΑ ΑΜΈΣΩΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Η ΛΕΙΤΟΥΡΓΊΑ ΤΟΥ ΣΥΜΠΥΚΝΩΤΉ
ΟΞΥΓΌΝΟΥ ΕΚΤΌΣ ΤΩΝ ΠΡΟΔΙΑΓΡΑΦΏΝ
ΘΕΡΜΟΚΡΑΣΊΑΣ ΛΕΙΤΟΥΡΓΊΑΣ ΜΠΟΡΕΊ ΝΑΠΕΡΙΟΡΊΣΕΙ ΤΗ ΔΥΝΑΤΌΤΗΤΑ ΤΟΥ ΣΥΜΠΥΚΝΩΤΉ ΝΑΙΚΑΝΟΠΟΙΕΊ ΤΗΝ ΠΡΟΔΙΑΓΡΑΦΉ ΤΟΥ ΣΥΜΠΥΚΝΩΤΉ
ΟΞΥΓΌΝΟΥ. ΑΝΑΤΡΈΞΤΕ ΣΤΗΝ ΕΝΌΤΗΤΑ
ΠΡΟΔΙΑΓΡΑΦΏΝ ΤΟΥ ΠΑΡΌΝΤΟΣ ΕΓΧΕΙΡΙΔΊΟΥ ΓΙΑΤΑ ΌΡΙΑ ΘΕΡΜΟΚΡΑΣΊΑΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΧΡΗΣΙΜΟΠΟΙΕΊΤΕ ΛΆΔΙ,ΓΡΆΣΟ Ή ΠΡΟΪΌΝΤΑ ΜΕ ΒΆΣΗ ΠΕΤΡΈΛΑΙΟ Ή ΆΛΛΑΕΎΦΛΕΚΤΑ ΠΡΟΪΌΝΤΑ ΜΕ ΤΑ ΠΑΡΕΛΚΌΜΕΝΑ
ΠΟΥ ΦΈΡΟΥΝ ΟΞΥΓΌΝΟ Ή ΜΕ ΤΟΝ ΣΥΜΠΥΚΝΩΤΉ
ΟΞΥΓΌΝΟΥ. ΠΡΈΠΕΙ ΝΑ ΧΡΗΣΙΜΟΠΟΙΟΎΝΤΑΙ ΜΌΝΟΛΟΣΙΌΝ Ή ΑΛΟΙΦΈΣ ΜΕ ΒΆΣΗ ΤΟ ΝΕΡΌ Ή ΣΥΜΒΑΤΈΣΜΕ ΤΟ ΟΞΥΓΌΝΟ. ΤΟ ΟΞΥΓΌΝΟ ΕΠΙΤΑΧΎΝΕΙ ΤΗΝΚΑΎΣΗ ΤΩΝ ΕΎΦΛΕΚΤΩΝ ΟΥΣΙΏΝ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Η ΛΑΝΘΑΣΜΈΝΗ ΧΡΉΣΗ ΤΗΣΜΠΑΤΑΡΊΑΣ ΜΠΟΡΕΊ ΝΑ ΠΡΟΚΑΛΈΣΕΙ ΘΈΡΜΑΝΣΗΉ ΑΝΆΦΛΕΞΗ ΤΗΣ ΜΠΑΤΑΡΊΑΣ ΚΑΙ ΜΠΟΡΕΊ ΝΑΠΡΟΚΑΛΈΣΕΙ ΣΟΒΑΡΌ ΤΡΑΥΜΑΤΙΣΜΌ. ΔΕΝΕΠΙΤΡΈΠΕΤΑΙ ΝΑ ΤΡΥΠΉΣΕΤΕ, ΝΑ ΧΤΥΠΉΣΕΤΕ,ΝΑ ΠΑΤΉΣΕΤΕ Ή ΝΑ ΡΊΞΕΤΕ ΤΗΝ ΜΠΑΤΑΡΊΑ,Ή ΝΑ ΥΠΟΒΆΛΕΤΕ ΑΛΛΙΏΣ ΤΗΝ ΜΠΑΤΑΡΊΑ ΣΕΙΣΧΥΡΈΣ ΚΡΟΎΣΕΙΣ Ή ΚΡΑΔΑΣΜΟΎΣ. Η ΧΡΉΣΗ
ΜΙΑΣ ΚΑΤΕΣΤΡΑΜΜΈΝΗΣ ΜΠΑΤΑΡΊΑΣ ΜΠΟΡΕΊ ΝΑΠΡΟΚΑΛΈΣΕΙ ΠΡΟΣΩΠΙΚΌ ΤΡΑΥΜΑΤΙΣΜΌ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΒΡΑΧΥΚΥΚΛΏΝΕΤΕ ΤΙΣΜΕΤΑΛΛΙΚΈΣ ΕΠΑΦΈΣ ΤΗΣ ΜΠΑΤΑΡΊΑΣ ΜΕΜΕΤΑΛΛΙΚΆ ΑΝΤΙΚΕΊΜΕΝΑ, ΌΠΩΣ ΚΛΕΙΔΙΆ ΉΚΈΡΜΑΤΑ. ΚΆΤΙ ΤΈΤΟΙΟ ΜΠΟΡΕΊ ΝΑ ΠΡΟΚΑΛΈΣΕΙ
ΣΠΙΝΘΉΡΕΣ Ή ΤΗΝ ΠΑΡΑΓΩΓΉ ΥΠΕΡΒΟΛΙΚΉΣ
ΘΕΡΜΌΤΗΤΑΣ.

FreeStyle ComfortΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΚΑΤΆ ΤΗ ΧΡΉΣΗ ΤΟΥ ΦΟΡΗΤΟΎ
ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ ΣΕ ΕΞΩΤΕΡΙΚΌ
ΧΏΡΟ ΜΕ ΤΟ ΤΡΟΦΟΔΟΤΙΚΌ AC, ΣΥΝΔΈΣΤΕ ΤΟΤΡΟΦΟΔΟΤΙΚΌ ΑΠΟΚΛΕΙΣΤΙΚΆ ΣΕ ΜΙΑ ΠΡΊΖΑ ΜΕΔΙΑΚΌΠΤΗ ΒΛΆΒΗΣ ΓΕΊΩΣΗΣ (GFI).
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Ο ΣΥΜΠΥΚΝΩΤΉΣ ΟΞΥΓΌΝΟΥ
ΔΕΝ ΠΡΈΠΕΙ ΝΑ ΧΡΗΣΙΜΟΠΟΙΕΊΤΑΙ ΠΑΡΑΚΕΊΜΕΝΑΣΕ Ή ΣΕ ΔΙΆΤΑΞΗ ΣΤΟΊΒΑΣ ΜΕ ΆΛΛΟ ΕΞΟΠΛΙΣΜΌ.
ΕΆΝ Η ΧΡΉΣΗ ΠΑΡΑΚΕΊΜΕΝΑ Ή ΣΕ ΔΙΆΤΑΞΗ
ΣΤΟΊΒΑΣ ΕΊΝΑΙ ΑΝΑΠΌΦΕΥΚΤΗ, Η ΣΥΣΚΕΥΉ ΠΡΈΠΕΙΝΑ ΠΑΡΑΚΟΛΟΥΘΕΊΤΑΙ ΓΙΑ ΝΑ ΕΠΑΛΗΘΕΎΕΤΑΙ ΗΚΑΝΟΝΙΚΉ ΛΕΙΤΟΥΡΓΊΑ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Η ΧΡΉΣΗ ΚΑΛΩΔΊΩΝ ΚΑΙΠΡΟΣΑΡΜΟΓΈΩΝ ΕΚΤΌΣ ΑΠΌ ΤΑ ΠΡΟΒΛΕΠΌΜΕΝΑ,ΜΕ ΤΗΝ ΕΞΑΊΡΕΣΗ ΤΩΝ ΚΑΛΩΔΊΩΝ ΚΑΙΠΡΟΣΑΡΜΟΓΈΩΝ ΠΟΥ ΠΩΛΟΎΝΤΑΙ ΑΠΌ ΤΟΝΚΑΤΑΣΚΕΥΑΣΤΉ ΤΟΥ ΙΑΤΡΙΚΟΎ ΗΛΕΚΤΡΙΚΟΎ
ΕΞΟΠΛΙΣΜΟΎ ΩΣ ΑΝΤΑΛΛΑΚΤΙΚΆ ΓΙΑ ΕΣΩΤΕΡΙΚΆ
ΕΞΑΡΤΉΜΑΤΑ, ΜΠΟΡΕΊ ΝΑ ΟΔΗΓΉΣΕΙ ΣΕΑΥΞΗΜΈΝΕΣ ΕΚΠΟΜΠΈΣ Ή ΜΕΙΩΜΈΝΗ ΑΤΡΩΣΊΑ ΤΟΥΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΧΡΗΣΙΜΟΠΟΙΕΊΤΕ ΜΌΝΟ ΤΗΝΗΛΕΚΤΡΙΚΉ ΤΆΣΗ ΠΟΥ ΚΑΘΟΡΊΖΕΤΑΙ ΣΤΗΝ ΕΤΙΚΈΤΑ
ΠΡΟΔΙΑΓΡΑΦΏΝ ΠΟΥ ΦΈΡΕΙ Η ΣΥΣΚΕΥΉ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΧΡΗΣΙΜΟΠΟΙΕΊΤΕ ΚΑΛΏΔΙΑ
ΕΠΈΚΤΑΣΗΣ ΜΕ ΑΥΤΉΝ ΤΗ ΜΟΝΆΔΑ ΚΑΙ ΜΗΣΥΝΔΈΕΤΕ ΥΠΕΡΒΟΛΙΚΌ ΑΡΙΘΜΌ ΒΥΣΜΆΤΩΝ ΣΤΗΝΊΔΙΑ ΠΑΡΟΧΉ ΡΕΎΜΑΤΟΣ. Η ΧΡΉΣΗ ΚΑΛΩΔΊΩΝ
ΕΠΈΚΤΑΣΗΣ ΘΑ ΜΠΟΡΟΎΣΕ ΝΑ ΕΠΗΡΕΆΣΕΙ
ΑΡΝΗΤΙΚΆ ΤΗΝ ΑΠΌΔΟΣΗ ΤΗΣ ΣΥΣΚΕΥΉΣ.
ΥΠΕΡΒΟΛΙΚΌΣ ΑΡΙΘΜΌΣ ΒΥΣΜΆΤΩΝ ΣΕ ΜΊΑΠΑΡΟΧΉ ΜΠΟΡΕΊ ΝΑ ΟΔΗΓΉΣΕΙ ΣΕ ΥΠΕΡΦΌΡΤΩΣΗ
ΤΟΥ ΗΛΕΚΤΡΙΚΟΎ ΠΊΝΑΚΑ ΠΡΟΚΑΛΏΝΤΑΣ ΤΗΝΕΝΕΡΓΟΠΟΊΗΣΗ ΤΟΥ ΑΣΦΑΛΕΙΟΔΙΑΚΌΠΤΗ/ ΤΗΣΑΣΦΆΛΕΙΑΣ, Ή ΠΥΡΚΑΓΙΆ ΣΕ ΠΕΡΊΠΤΩΣΗ ΠΟΥΟ ΑΣΦΑΛΕΙΟΔΙΑΚΌΠΤΗΣ/ Η ΑΣΦΆΛΕΙΑ ΔΕΝΛΕΙΤΟΥΡΓΉΣΟΥΝ.
ΠΡΟΣΟΧΗ: Στην περίπτωση που εκδηλωθεί
συναγερμός ή που παρατηρήσετε ότι οσυμπυκνωτής οξυγόνου δεν λειτουργεί
κανονικά, συμβουλευτείτε την ενότητα
«Αντιμετώπιση προβλημάτων» στο παρόν
εγχειρίδιο. Εάν δεν μπορείτε να επιλύσετε
το πρόβλημα, συμβουλευτείτε τον πάροχο
του εξοπλισμού σας.

ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗΝ ΕΠΙΧΕΙΡΉΣΕΤΕ
ΟΠΟΙΑΔΉΠΟΤΕ ΕΡΓΑΣΊΑ ΣΥΝΤΉΡΗΣΗΣ
ΕΚΤΌΣ ΑΠΌ ΤΙΣ ΠΙΘΑΝΈΣ ΛΎΣΕΙΣ ΠΟΥΠΑΡΑΤΊΘΕΝΤΑΙ ΣΤΟ ΠΑΡΌΝ ΕΓΧΕΙΡΊΔΙΟ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΧΡΗΣΙΜΟΠΟΙΕΊΤΕ ΜΌΝΟΠΑΡΕΛΚΌΜΕΝΑ ΠΟΥ ΣΥΝΙΣΤΏΝΤΑΙ ΑΠΌ ΤΟΝΚΑΤΑΣΚΕΥΑΣΤΉ. Η ΧΡΉΣΗ ΟΠΟΙΩΝΔΉΠΟΤΕ ΆΛΛΩΝ
ΜΠΟΡΕΊ ΝΑ ΕΊΝΑΙ ΕΠΙΚΊΝΔΥΝΗ, ΝΑ ΠΡΟΚΑΛΈΣΕΙ
ΣΟΒΑΡΉ ΖΗΜΙΆ ΣΤΟΝ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ ΚΑΙΝΑ ΚΑΤΑΣΤΉΣΕΙ ΤΗΝ ΕΓΓΎΗΣΗ ΜΗ ΈΓΚΥΡΗ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Ο ΚΑΤΑΣΚΕΥΑΣΤΉΣ ΣΥΝΙΣΤΆ
ΜΙΑ ΕΝΑΛΛΑΚΤΙΚΉ ΠΗΓΉ ΣΥΜΠΛΗΡΩΜΑΤΙΚΟΎ
ΟΞΥΓΌΝΟΥ ΣΕ ΠΕΡΊΠΤΩΣΗ ΔΙΑΚΟΠΉΣ ΡΕΎΜΑΤΟΣ,ΚΑΤΆΣΤΑΣΗΣ ΣΥΝΑΓΕΡΜΟΎ Ή ΜΗΧΑΝΙΚΉΣ ΒΛΆΒΗΣ.
ΣΥΜΒΟΥΛΕΥΤΕΊΤΕ ΤΟΝ ΓΙΑΤΡΌ Ή ΤΟΝ ΠΆΡΟΧΟ
ΤΟΥ ΕΞΟΠΛΙΣΜΟΎ ΓΙΑ ΤΟΝ ΤΎΠΟ ΤΟΥ ΕΦΕΔΡΙΚΟΎ
ΣΥΣΤΉΜΑΤΟΣ ΠΟΥ ΑΠΑΙΤΕΊΤΑΙ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗΝ ΑΦΉΝΕΤΕ ΝΑ ΦΡΆΞΟΥΝΤΑ ΑΝΟΊΓΜΑΤΑ ΕΙΣΑΓΩΓΉΣ Ή ΕΞΌΔΟΥ ΑΈΡΑ.
ΜΗΝ ΑΦΉΝΕΤΕ ΝΑ ΠΈΣΟΥΝ ΚΑΙ ΜΗΝ ΕΙΣΆΓΕΤΕ
ΑΝΤΙΚΕΊΜΕΝΑ ΣΕ ΚΑΝΈΝΑ ΆΝΟΙΓΜΑ ΤΗΣΣΥΣΚΕΥΉΣ. ΚΆΤΙ ΤΈΤΟΙΟ ΜΠΟΡΕΊ ΝΑ ΠΡΟΚΑΛΈΣΕΙ
ΥΠΕΡΘΈΡΜΑΝΣΗ ΤΟΥ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ ΚΑΙΝΑ ΕΠΗΡΕΆΣΕΙ ΑΡΝΗΤΙΚΆ ΤΗΝ ΑΠΌΔΟΣΗ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΤΟ ΚΆΠΝΙΣΜΑ ΕΝΏΧΡΗΣΙΜΟΠΟΙΕΊΤΑΙ ΟΞΥΓΌΝΟ ΕΊΝΑΙ Η ΠΡΏΤΗ
ΑΙΤΊΑ ΤΡΑΥΜΑΤΙΣΜΏΝ ΑΠΌ ΠΥΡΚΑΓΙΆ ΚΑΙΣΧΕΤΙΖΌΜΕΝΩΝ ΘΑΝΆΤΩΝ. ΠΡΈΠΕΙ ΝΑ ΤΗΡΕΊΤΕ ΤΙΣΑΚΌΛΟΥΘΕΣ ΠΡΟΕΙΔΟΠΟΙΉΣΕΙΣ ΑΣΦΑΛΕΊΑΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗΝ ΕΠΙΤΡΈΠΕΤΕ ΤΟ ΚΆΠΝΙΣΜΑ,ΤΑ ΚΕΡΙΆ Ή ΤΙΣ ΑΝΟΙΧΤΈΣ ΦΛΌΓΕΣ ΣΤΟΝ ΊΔΙΟ ΧΏΡΟΜΕ ΤΗ ΣΥΣΚΕΥΉ Ή ΤΑ ΠΑΡΕΛΚΌΜΕΝΑ ΠΟΥ ΦΈΡΟΥΝ
ΟΞΥΓΌΝΟ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΤΟ ΚΆΠΝΙΣΜΑ ΕΝΏ ΦΟΡΆΤΕ
ΚΆΝΟΥΛΑ ΟΞΥΓΌΝΟΥ ΜΠΟΡΕΊ ΝΑ ΠΡΟΚΑΛΈΣΕΙ
ΕΓΚΑΎΜΑΤΑ ΤΟΥ ΠΡΟΣΏΠΟΥ ΚΑΙ ΠΙΘΑΝΏΣ
ΘΆΝΑΤΟ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΕΆΝ ΚΑΠΝΊΖΕΤΕ, ΤΑ ΑΚΌΛΟΥΘΑ
3 ΒΉΜΑΤΑ ΜΠΟΡΕΊ ΝΑ ΣΏΣΟΥΝ ΤΗ ΖΩΉ ΣΑΣ:ΑΠΕΝΕΡΓΟΠΟΙΉΣΤΕ ΤΟΝ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ,ΑΦΑΙΡΈΣΤΕ ΤΗΝ ΚΆΝΟΥΛΑ ΚΑΙ ΒΓΕΊΤΕ ΑΠΌ ΤΟΝΧΏΡΟ ΌΠΟΥ ΒΡΊΣΚΕΤΑΙ Η ΣΥΣΚΕΥΉ.

ΠΡΟΣΟΧΗ: Προκειμένου να αποφύγετε την ακύρωση
της εγγύησης, ακολουθείτε όλες τις οδηγίες τουκατασκευαστή.

PN MN232-C4 L | Εγχειρίδιο χρήστη GR - 383

FreeStyle ComfortΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΤΑ ΣΉΜΑΤΑ «ΑΠΑΓΟΡΕΎΕΤΑΙΤΟ ΚΆΠΝΙΣΜΑ – ΧΡΉΣΗ ΟΞΥΓΌΝΟΥ» ΠΡΈΠΕΙ ΝΑΑΝΑΡΤΏΝΤΑΙ ΣΕ ΠΕΡΊΟΠΤΗ ΘΈΣΗ ΣΤΟ ΣΠΊΤΙ ΉΣΤΟΝ ΧΏΡΟ ΌΠΟΥ ΧΡΗΣΙΜΟΠΟΙΕΊΤΑΙ ΟΞΥΓΌΝΟ.
ΟΙ ΧΡΗΣΤΕΣ ΚΑΙ ΟΙ ΦΡΟΝΤΙΣΤΈΣ ΤΟΥΣ ΠΡΈΠΕΙ ΝΑΕΝΗΜΕΡΏΝΟΝΤΑΙ ΣΧΕΤΙΚΆ ΜΕ ΤΟΥΣ ΚΙΝΔΎΝΟΥΣ
ΤΟΥ ΚΑΠΝΊΣΜΑΤΟΣ ΥΠΌ ΤΗΝ ΠΑΡΟΥΣΊΑ, Ή ΚΑΤΆΤΗ ΔΙΆΡΚΕΙΑ ΤΗΣ ΧΡΉΣΗΣ, ΙΑΤΡΙΚΟΎ ΟΞΥΓΌΝΟΥ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΧΡΗΣΙΜΟΠΟΙΕΊΤΕ ΤΟΝΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ ΥΠΌ ΤΗΝ ΠΑΡΟΥΣΊΑ
ΕΎΦΛΕΚΤΩΝ ΑΕΡΊΩΝ. ΑΥΤΌ ΜΠΟΡΕΊ ΝΑ ΟΔΗΓΉΣΕΙΣΕ ΤΑΧΕΊΑ ΚΑΎΣΗ ΠΡΟΚΑΛΏΝΤΑΣ ΚΑΤΑΣΤΡΟΦΉ
ΙΔΙΟΚΤΗΣΊΑΣ, ΣΩΜΑΤΙΚΟΎΣ ΤΡΑΥΜΑΤΙΣΜΟΎΣΉ ΘΆΝΑΤΟ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Η ΑΦΑΊΡΕΣΗ ΤΗΣ ΚΆΝΟΥΛΑΣΚΑΙ Η ΑΠΌΘΕΣΉ ΤΗΣ ΕΠΆΝΩ ΣΕ ΕΝΔΎΜΑΤΑ,ΚΛΙΝΟΣΚΕΠΆΣΜΑΤΑ, ΚΑΝΑΠΈΔΕΣ, ΜΑΞΙΛΆΡΙΑ ΉΠΑΡΌΜΟΙΑ ΥΛΙΚΆ ΘΑ ΠΡΟΚΑΛΈΣΕΙ ΚΑΤΆΚΑΥΣΗ
ΑΕΡΊΟΥ ΝΈΦΟΥΣ ΌΤΑΝ ΕΚΤΕΘΕΊ ΣΕ ΤΣΙΓΆΡΟ,ΠΗΓΉ ΘΕΡΜΌΤΗΤΑΣ Ή ΦΛΌΓΑ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗΝ ΑΦΉΝΕΤΕ ΤΗ ΡΙΝΙΚΉ
ΚΆΝΟΥΛΑ ΕΠΆΝΩ Ή ΚΆΤΩ ΑΠΌ ΕΝΔΎΜΑΤΑ,ΚΛΙΝΟΣΚΕΠΆΣΜΑΤΑ Ή ΜΑΞΙΛΆΡΙΑ ΚΑΡΈΚΛΑΣ.
ΕΆΝ Η ΜΟΝΆΔΑ ΕΝΕΡΓΟΠΟΙΗΘΕΊ ΑΛΛΆ ΔΕΝΧΡΗΣΙΜΟΠΟΙΗΘΕΊ, ΤΟ ΟΞΥΓΌΝΟ ΘΑ ΚΑΤΑΣΤΉΣΕΙΤΟ ΥΛΙΚΌ ΕΎΦΛΕΚΤΟ. ΔΙΑΣΦΑΛΊΣΤΕ ΌΤΙ Η ΙΣΧΎΣ
ΕΊΝΑΙ ΑΠΕΝΕΡΓΟΠΟΙΗΜΕΝΗ ΌΤΑΝ Ο ΣΥΜΠΥΚΝΩΤΉΣ
ΟΞΥΓΌΝΟΥ ΔΕΝ ΒΡΊΣΚΕΤΑΙ ΣΕ ΧΡΉΣΗ.
ΠΡΟΕΙΔΟΠΟΊΗΣΗ: ΤΑ ΕΓΚΕΚΡΙΜΈΝΑ ΠΑΡΕΛΚΌΜΕΝΑ
ΠΑΡΟΧΉΣ ΟΞΥΓΌΝΟΥ ΈΧΟΥΝ ΣΧΕΔΙΑΣΤΕΊ ΓΙΑΝΑ ΜΕΙΏΝΟΥΝ ΤΗ ΔΙΆΔΟΣΗ ΤΗΣ ΦΩΤΙΆΣ ΓΙΑΤΗΝ ΑΣΦΆΛΕΙΑ ΤΟΥ ΑΣΘΕΝΟΎΣ. Η ΧΡΉΣΗ ΜΗΕΓΚΕΚΡΙΜΈΝΩΝ ΠΑΡΕΛΚΟΜΈΝΩΝ ΜΠΟΡΕΊ ΝΑΑΥΞΉΣΕΙ ΤΟΝ ΚΊΝΔΥΝΟ ΤΡΑΥΜΑΤΙΣΜΟΎ Ή ΖΗΜΙΏΝ
ΠΟΥ ΣΧΕΤΊΖΟΝΤΑΙ ΜΕ ΠΥΡΚΑΓΙΈΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Ο ΣΥΜΠΥΚΝΩΤΗΣ ΟΞΥΓΟΝΟΥΠΡΕΠΕΙ ΝΑ ΑΠΕΝΕΡΓΟΠΟΙΗΘΕΙ ΚΑΙ ΝΑΑΠΟΣΥΝΔΕΘΕΙ ΑΠΟ ΕΞΩΤΕΡΙΚΗ ΠΗΓΗ ΤΡΟΦΟΔΟΣΙΑΣΠΡΙΝ ΤΗΝ ΑΦΑΙΡΕΣΗ ΤΗΣ ΣΥΣΤΟΙΧΙΑΣ ΜΠΑΤΑΡΙΩΝ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΤΑ ΜΙΚΡΆ ΜΈΡΗΜΠΟΡΕΊ ΝΑ ΠΑΡΟΥΣΙΆΖΟΥΝ ΠΙΘΑΝΌ
ΚΊΝΔΥΝΟ ΠΝΙΓΜΟΎ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΑΠΟΦΎΓΕΤΕ ΤΗΝ ΕΠΑΦΉ ΜΕΤΟ ΆΝΟΙΓΜΑ ΑΠΑΓΌΜΕΝΟΥ ΑΈΡΑ ΓΙΑ ΠΕΡΙΣΣΌΤΕΡΟ
ΑΠΌ 10 ΔΕΥΤΕΡΌΛΕΠΤΑ ΚΑΘΏΣ ΜΠΟΡΕΊ ΝΑ ΕΊΝΑΙ
ΘΕΡΜΌ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Ο ΆΝΕΜΟΣ Ή ΤΑ ΙΣΧΥΡΆ ΡΕΎΜΑΤΑ
ΜΠΟΡΕΊ ΝΑ ΕΠΗΡΕΆΣΟΥΝ ΑΡΝΗΤΙΚΆ ΤΗΝ ΑΚΡΙΒΉ
ΧΟΡΉΓΗΣΗ ΟΞΥΓΟΝΟΘΕΡΑΠΕΊΑΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΑΥΤΌ ΤΟ ΠΡΟΪΌΝ ΜΠΟΡΕΊΝΑ ΣΑΣ ΕΚΘΈΣΕΙ ΣΕ ΧΗΜΙΚΈΣ ΟΥΣΊΕΣ
ΣΥΜΠΕΡΙΛΑΜΒΑΝΟΜΈΝΟΥ ΤΟΥ ΝΙΚΕΛΊΟΥ,ΤΟ ΟΠΟΊΟ ΕΊΝΑΙ ΓΝΩΣΤΌ ΣΤΗΝ ΠΟΛΙΤΕΊΑ ΤΗΣΚΑΛΙΦΌΡΝΙΑ ΌΤΙ ΠΡΟΚΑΛΕΊ ΚΑΡΚΊΝΟ. ΓΙΑΠΕΡΙΣΣΌΤΕΡΕΣ ΠΛΗΡΟΦΟΡΊΕΣ, ΜΕΤΑΒΕΊΤΕ ΣΤΟ
WWW.P65WARNINGS.CA.GOV.

384 - GR PN MN232-C4 L | Εγχειρίδιο χρήστη

ΠΡΟΣΟΧΗ: Αποσυνδέετε πάντα τοτροφοδοτικό AC από τον τοίχο πριν τηναποσύνδεση του τροφοδοτικού AC απότον συμπυκνωτή οξυγόνου.
ΠΡΟΣΟΧΗ: Μην τοποθετείτε τον συμπυκνωτή
οξυγόνου σε θέση που καθιστά δύσκολη τηνπρόσβαση στο καλώδιο ρεύματος.
ΠΡΟΣΟΧΗ: Τοποθετείτε πάντα τη σωλήνωση παροχής
οξυγόνου και τα καλώδια ρεύματος με τρόπο πουαποτρέπει τον κίνδυνο παραπατήματος ή τηνπιθανότητα τυχαίου στραγγαλισμού.
ΠΡΟΣΟΧΗ: Όταν χρησιμοποιείτε τον συμπυκνωτή
οξυγόνου σε αυτοκίνητο, σκάφος ή σε άλλες πηγές DCμε το τροφοδοτικό DC, βεβαιωθείτε ότι το όχημα έχειξεκινήσει και κινείται πριν συνδέσετε τον συμπυκνωτή
οξυγόνου. Η μη τήρηση αυτών των οδηγιών μπορεί ναέχει ως αποτέλεσμα την αποτυχία του τροφοδοτικού
να τροφοδοτήσει με ισχύ τον συμπυκνωτή οξυγόνου.
ΠΡΟΣΟΧΗ: Όταν το όχημα στο οποίο χρησιμοποιείτε τημονάδα συμπυκνωτή οξυγόνου είναι απενεργοποιημένο,αποσυνδέστε και αφαιρέστε τη συσκευή από το όχημα.
Μην αποθηκεύετε τον συμπυκνωτή οξυγόνου σε έναπολύ θερμό ή ψυχρό όχημα ή σε άλλο παρόμοιο
περιβάλλον με υψηλή ή χαμηλή θερμοκρασία.
Σημείωση: Φορητός και κινητός εξοπλισμός
επικοινωνιών με ραδιοσυχνότητες (RF) μπορεί ναεπηρεάσει τον ιατρικό ηλεκτρικό εξοπλισμό.
ΠΡΟΣΟΧΗ: ΜΗΝ αφήνετε τον συμπυκνωτή
οξυγόνου ή το τροφοδοτικό συνδεδεμένοστο όχημα, εάν η ανάφλεξη βρίσκεται στηθέση ΑΠΕΝΕΡΓΟΠΟΙΗΣΗΣ. Αυτό μπορεί ναεξαντλήσει την μπαταρία του οχήματος.
ΠΡΟΣΟΧΗ: Η λειτουργία ή αποθήκευση τουσυμπυκνωτή οξυγόνου εκτός του κανονικού εύρους
θερμοκρασίας λειτουργίας και πίεσης περιβάλλοντος
(υψόμετρο) μπορεί να επηρεάσει την απόδοση καινα μειώσει τη διάρκεια ζωής της μπαταρίας ή/και νααυξήσει τον χρόνο φόρτισης της μπαταρίας. Ανατρέξτε
στην ενότητα «Προδιαγραφές» στο παρόν εγχειρίδιο
για τα όρια θερμοκρασίας αποθήκευσης και λειτουργίας.
ΠΡΟΣΟΧΗ: Για συμπυκνωτές οξυγόνου που είναι
εξοπλισμένοι με μπαταρίες: Αποθηκεύστε σε δροσερό
και ξηρό μέρος για να βοηθήσετε να διασφαλιστεί
η μακροβιότητα της μπαταρίας. Η αποθήκευση τουσυμπυκνωτή οξυγόνου για παρατεταμένες χρονικές
περιόδους σε υψηλές θερμοκρασίες ή με πλήρως
φορτισμένη ή πλήρως αποφορτισμένη μπαταρία
μπορεί να υποβαθμίσει τη συνολική διάρκεια ζωήςτης μπαταρίας. Μην επιχειρήσετε να ανοίξετε τηνμπαταρία, καθώς δεν υπάρχουν μέρη που επιδέχονται
συντήρηση στο εσωτερικό της μπαταρίας. Φυλάσσετε
τις μπαταρίες μακριά από παιδιά.
ΠΡΟΣΟΧΗ: ΧΡΗΣΙΜΟΠΟΙΕΙΤΕ ΜΟΝΟ τις μπαταρίες
που παρέχονται από τον κατασκευαστή. Για σωστή
απόρριψη, επικοινωνήστε με τον πάροχο τουεξοπλισμού σας ή με την αρμόδια υπηρεσία τοπικής
αυτοδιοίκησης για τις απαιτήσεις απόρριψης.

FreeStyle ComfortΣημείωση: Για να μετριάσει τη βλάβη που προκαλείται
από αστοχία του συμπιεστή ή διακοπή ρεύματος, οπρομηθευτής του εξοπλισμού θα πρέπει να αξιολογεί τηνανάγκη για εφεδρικό οξυγόνο κατά την εγκατάσταση καιπεριοδικά όταν τα παρακάτω χαρακτηριστικά αλλάζουν:- Η κατάσταση του ασθενούς.
- Το περιβάλλον στο οποίο ζει ο ασθενής.
- Η δυνατότητα παροχής εκ νέου στον ασθενή εφεδρικών
προμηθειών συμπληρωματικού οξυγόνου.
Σημείωση: Διασφαλίστε ότι η κάνουλα έχει εισαχθεί καιστερεωθεί πλήρως. Αυτό διασφαλίζει ότι ο συμπυκνωτής
οξυγόνου μπορεί να ανιχνεύει σωστά την εισπνοή γιαχορήγηση οξυγόνου. Κατά τη διάρκεια της εισπνοής,θα πρέπει να ακούσετε ή να αισθανθείτε τη ροή τουοξυγόνου στα άκρα της ρινικής κάνουλας.

Σημείωση: Εάν ο συμπυκνωτής οξυγόνου έχειαποθηκευτεί για παρατεταμένη χρονική περίοδο εκτός
των κανονικών περιβαλλοντικών συνθηκών λειτουργίας
του, όπως φαίνονται στον πίνακα προδιαγραφών,θα πρέπει να αφεθεί να επιστρέψει στην κανονική
θερμοκρασία λειτουργίας προτού ενεργοποιηθεί.
Σημείωση: Ο προμηθευτής του εξοπλισμού θα πρέπει
να αξιολογεί την ανάγκη για εφεδρικό οξυγόνο σεπερίπτωση αστοχίας του συμπιεστή ή διακοπής
ρεύματος κατά την εγκατάσταση και περιοδικά, όταντα παρακάτω χαρακτηριστικά αλλάζουν:- Η κατάσταση του ασθενούς.
- Το περιβάλλον στο οποίο ζει ο ασθενής.
- Η δυνατότητα παροχής εκ νέου στον ασθενή εφεδρικών
προμηθειών συμπληρωματικού οξυγόνου.

Σημείωση: Οι μπαταρίες λιθίου μπορεί να χάσουν μόνιματη χωρητικότητά τους όταν εκτεθούν σε υπερβολικά
υψηλές θερμοκρασίες με τις μπαταρίες τελείως
φορτισμένες ή τελείως εξαντλημένες. Για μακροχρόνια
αποθήκευση, συνιστάται οι μπαταρίες να είναι
φορτισμένες κατά 25 έως 50% και να παραμένουν εντός
του εύρους θερμοκρασίας -20°C–20°C (-4°F–68°F).
Σημείωση: Μη λειτουργείτε τον συμπυκνωτή οξυγόνου
χωρίς να είναι τοποθετημένο το φίλτρο εισαγωγής αέρα.
Εάν παρέχεται ένα δεύτερο φίλτρο, τοποθετήστε τοανταλλακτικό φίλτρο πριν να καθαρίσετε το βρώμικο
φίλτρο. Καθαρίστε το βρώμικο φίλτρο με ζεστό διάλυμα
σαπουνιού και νερού και στη συνέχεια στεγνώστε
τελείως πριν από τη χρήση.
Σημείωση: Ο κατασκευαστής δεν συνιστά τηναποστείρωση αυτού του εξοπλισμού.
Σημείωση: Δεν υπάρχει ποτέ κίνδυνος εξάντλησης τουοξυγόνου σε ένα δωμάτιο όταν χρησιμοποιείτε τη μονάδα
συμπυκνωτή οξυγόνου.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Ο ΣΥΜΠΥΚΝΩΤΉΣ
ΟΞΥΓΌΝΟΥ ΔΕΝ ΠΡΈΠΕΙ ΝΑ ΛΕΙΤΟΥΡΓΕΊΉ ΝΑ ΣΥΝΔΈΕΤΑΙ ΣΕ ΕΞΩΤΕΡΙΚΉ ΙΣΧΎ(AC/DC) ΧΩΡΊΣ ΤΟΠΟΘΕΤΗΜΈΝΗ
ΜΠΑΤΑΡΊΑ. ΜΗΝ ΕΠΙΧΕΙΡΉΣΕΤΕ ΝΑΕΝΕΡΓΟΠΟΙΉΣΕΤΕ ΤΗ ΜΟΝΆΔΑ Ή ΝΑ ΤΗΣΥΝΔΈΣΕΤΕ ΣΕ ΕΞΩΤΕΡΙΚΉ ΠΗΓΉ ΙΣΧΎΟΣ
ΌΤΑΝ ΤΟ ΠΑΚΈΤΟ ΜΠΑΤΑΡΊΑΣ ΈΧΕΙΑΦΑΙΡΕΘΕΊ.
ΑΦΑΙΡΈΣΤΕ ΤΟ ΠΑΚΈΤΟ ΜΠΑΤΑΡΊΑΣ ΜΌΝΟΌΤΑΝ Η ΜΟΝΆΔΑ ΕΊΝΑΙ ΑΠΕΝΕΡΓΟΠΟΙΗΜΕΝΗΚΑΙ ΑΠΟΣΥΝΔΕΔΕΜΈΝΗ ΑΠΌ ΤΗΝ ΤΡΟΦΟΔΟΣΊΑ
ΡΕΎΜΑΤΟΣ. ΜΗΝ ΕΠΙΧΕΙΡΉΣΕΤΕ ΝΑ ΑΦΑΙΡΈΣΕΤΕΤΟ ΠΑΚΈΤΟ ΜΠΑΤΑΡΊΑΣ ΕΝΏ Η ΜΟΝΆΔΑ ΕΊΝΑΙ
ΑΚΌΜΑ ΕΝΕΡΓΟΠΟΙΗΜΕΝΗ Ή ΣΥΝΔΕΔΕΜΈΝΗ ΣΤΗΝΤΡΟΦΟΔΟΣΊΑ ΡΕΎΜΑΤΟΣ.

PN MN232-C4 L | Εγχειρίδιο χρήστη GR - 385

FreeStyle ComfortΠροδιαγραφές
FreeStyle ComfortΔιαστάσεις (με μονή μπαταρία)25,4 x 18,5 x 7,9 cm (10,0 x 7,3 x 3,1 in)Διαστάσεις (με διπλή μπαταρία)27,9 x 18,5 x 7,9 cm (11,0 x 7,3 x 3,1 in)Βάρος (με πακέτο μονής μπαταρίας)2,3 kg (5 lb)Βάρος (με πακέτο διπλής μπαταρίας)2,7 kg (6 lb)Ονομαστικό επίπεδο ήχου***Ρυθμίσεις ροήςΡύθμιση εξόδου O2([@20bpm STPD] Όλοι οι όγκοι +/- 15%)Συμπύκνωση οξυγόνου*Σύστημα παρακολούθησης οξυγόνου
Μέγιστη πίεση εξόδου
Ευαισθησία δοσολόγησης

39,93 dB(A) ± 0,42 dB(A) @ 2Ρύθμιση παλμού 1, 2, 3, 4, 5Ρύθμιση 1 - 210 mL/minΡύθμιση 2 - 420 mL/minΡύθμιση 3 - 630 mL/minΡύθμιση 4 - 840 mL/minΡύθμιση 5 - 1050 mL/min
90% (+5,5% / -3%)Ναι<30 PSI> -0,1 cm H2O (ρυθμιζόμενη)Ισχύς AC

100–240 VAC, 50–60 Hz, 120W μέγ.

Ισχύς DC

11–18 VDC (10 amp μέγ.)Τύπος μπαταρίας Ιόν λιθίου
Χωρητικότητα μπαταρίας Μονό στοιχείο: 6700 mAh, Διπλό στοιχείο: 13400 mAhΧρόνος επαναφόρτισης μπαταρίας

Μονή μπαταρία: 3,5 ώρεςΔιπλή μπαταρία: 6,0 ώρεςΔιάρκεια πακέτου μπαταρίας

Διάρκεια μονής μπαταρίας: Ρύθμιση 2 - έως 4 ώρεςΔιάρκεια διπλής μπαταρίας: Ρύθμιση 2 - έως 8 ώρεςΘερμοκρασία λειτουργίας**5°C έως 40°C (41°F έως 104°F)

Θερμοκρ. μεταφοράς και αποθήκευσης
(μονάδα)-25°C έως +70°C (-13°F έως +158°F) έως και τα 3048 m (10.000 ft)

Θερμοκρ. μεταφοράς και αποθήκευσης
(μπαταρία)-25°C έως +70°C (-13°F έως +158°F) έως και τα 3048 m (10.000 ft)

Υψόμετρο λειτουργίας**Υγρασία λειτουργίας

-381 έως 3048 m (-1250 έως 10.000 ft)(ελέγχθηκε στα 700–1060 hPa)
15–95% σχετική υγρασία (χωρίς συμπύκνωση υδρατμών)Υγρασία αποθήκευσης

0–90% χωρίς συμπύκνωση υδρατμών

Περιορισμένη εγγύηση

Δείτε τη δήλωση εγγύησης

* Με βάση ατμοσφαιρική πίεση 101 kPa (14,7 psi) στους 21°C (70°F)** Η μονάδα θα λειτουργεί σε όλα τα περιβάλλοντα, εφόσον το περιβάλλον βρίσκεται εντός του προσδιορισμένου
εύρους θερμοκρασίας και υγρασίας. Η λειτουργία εκτός αυτών των προδιαγραφών λειτουργίας μπορεί να περιορίσει τηδυνατότητα του συμπυκνωτή να ικανοποιεί την προδιαγραφή του συμπυκνωτή οξυγόνου σε υψηλότερους ρυθμούς ροήςλίτρων. Η οθόνη προθέρμανσης μπορεί να συνεχίσει να εμφανίζεται μέχρι ο συμπυκνωτής να φθάσει τη συγκέντρωσηστόχο ή έως και 15 λεπτά. Συνεχίζετε να χρησιμοποιείτε τη μονάδα κατά τη διάρκεια αυτού του χρονικού διαστήματος.
*** Η στάθμη ήχου μετρήθηκε σύμφωνα με τη μέθοδο δοκιμής με αρ. 14-1 10/2018 MDS-Hi.
Η ελάχιστη αναμενόμενη ωφέλιμη διάρκεια ζωής είναι πέντε χρόνια.

386 - GR PN MN232-C4 L | Εγχειρίδιο χρήστη

FreeStyle ComfortΟδηγίες λειτουργίας
1. Καθορίστε τη θέση και τοποθετήστε το FreeStyle
Comfort σε μια κατάλληλη θέση για χρήση.
α. Ο συμπυκνωτής πρέπει να αερίζεται καλά έτσιώστε τα φίλτρα εισαγωγής αέρα και η έξοδος
αέρα να μην εμποδίζονται.
β. Ο συμπυκνωτής πρέπει να τοποθετείται μετέτοιον τρόπο ώστε όλες οι ηχητικές και οπτικές
ενδείξεις και συναγερμοί να είναι εύκολα
ακουστοί και ορατοί.
γ. Βεβαιωθείτε ότι τα φίλτρα εισαγωγής αέραβρίσκονται στη θέση τους πριν τη λειτουργία
της μονάδας. Εάν ένα φίλτρο λείπει από τημονάδα, επικοινωνήστε με τον πάροχο τουεξοπλισμού σας.

3. Εισάγετε το τροφοδοτικό AC ή DC μέσα στηνυποδοχή θύρας τροφοδοσίας στην πρόσοψη, στηνκάτω δεξιά πλευρά της μονάδας.

2. Εγκαταστήστε ένα πακέτο μπαταρίας στον
συμπυκνωτή.
α. Ευθυγραμμίστε την μπαταρία με το κάτωπερίβλημα από το κέντρο του FreeStyle
Comfort.

4. Συνδέστε μια ρινική κάνουλα στη θύρα εξόδου
οξυγόνου. Διασφαλίστε ότι η σύνδεση είναι
ασφαλής πριν τη λειτουργία.
ΣΗΜΕΙΩΣΗ: Η χρήση διαφορετικής(ών) κάνουλας(ών)από τη συνιστώμενη μπορεί να επηρεάσει την απόδοση.

Σημείωση: Προς τον πάροχο του εξοπλισμού: Ταακόλουθα παρελκόμενα χορήγησης οξυγόνου
συνιστώνται για χρήση με το FreeStyle Comfort:β. Ολισθήστε την μπαταρία προς τα μέσα μέχρι ναακούσετε ένα κλικ για να επιβεβαιώσετε ότι έχεικλειδώσει στη θέση της.

• Ρινική κάνουλα με σωλήνωση 2,1 m (7 ft) (μέγιστη τιμήLPM): 6 LPM): Κωδικός εξαρτήματος CAIRE 5408-SEQ• Επιπρόσθετες πληροφορίες σχετικά με τα συνιστώμενα
παρελκόμενα είναι διαθέσιμες στο διαδίκτυο στηδιεύθυνση www.caireinc.com.
5. Ενεργοποιήστε το FreeStyle Comfort πατώνταςτο κουμπί λειτουργίας και επιλέξτε τησυνταγογραφημένη ρύθμιση ροής χρησιμοποιώντας
τα κουμπιά αύξησης (+) ή μείωσης (-) ροής.

PN MN232-C4 L | Εγχειρίδιο χρήστη GR - 387

FreeStyle ComfortΑναπνευστικός
ρυθμός
15
20
25
30
35
40
Θα παραχθεί ένας σύντομος ηχητικός συναγερμός
υποδεικνύοντας ότι η συσκευή έχει ενεργοποιηθεί.
Σημείωση: Όταν το FreeStyle Comfort ενεργοποιείται
για πρώτη φορά, η προεπιλεγμένη ρύθμιση ροής είναι 2.
Εάν η ρύθμιση αλλάξει, η μονάδα θα ενεργοποιηθεί στηνπροηγούμενη ρύθμιση ροής.
6. Τοποθετήστε τη ρινική κάνουλα στο πρόσωπό
σας σύμφωνα με τις οδηγίες του κατασκευαστή
που παρέχονται με την κάνουλα και αναπνέετε
κανονικά. Θα πρέπει να αισθανθείτε την χορήγηση
μιας ριπής οξυγόνου διαμέσου της ρινικής
κάνουλας κάθε φορά που εισπνέετε.
7. Για να αλλάξετε τη ρύθμιση ροής, πατήστε τακουμπιά αύξησης (+) ή μείωσης (-) ροής μέχρι
τον επιθυμητό ρυθμό ροής. Είναι φυσιολογικό ναακούσετε μια διαφορά στον ήχο καθώς αλλάζετετις ρυθμίσεις ροής.

Ρύθμιση όγκου παλμού μη-autoSat (mL)*
1
2
3
4
5
14,0
10,5
8,4
7,0
6,0
5,3

28,0
21,0
16,8
14,0
12,0
10,5

42,0
31,5
25,2
21,0
18,0
15,8

56,0
42,0
33,6
28,0
24,0
21,0

70,0
52,5
42,0
35,0
30,0
26,3

8. Απενεργοποιήστε το FreeStyle Comfort πατώνταςτο κουμπί λειτουργίας.
Σημείωση: Η οθόνη ενδείξεων θα τεθεί σε κατάσταση
αναμονής και θα σκοτεινιάσει μετά από 20 δευτερόλεπτα.
Πατήστε οποιοδήποτε κουμπί για να ενεργοποιήσετε τηνοθόνη από την κατάσταση αναμονής.
9. Για προηγμένα χαρακτηριστικά, βλ.
Οδηγίες πλοήγησης μενού

ΕΙΔΙΚΆ ΧΑΡΑΚΤΗΡΙΣΤΙΚΆ ΠΡΟΪΌΝΤΟΣ
autoSAT®
Η ιδιόκτητη λειτουργία autoSAT της CAIRE επιτρέπειστο FreeStyle Comfort να παρέχει σταθερό όγκοοξυγόνου (bolus) ανά αναπνοή, μέχρι ένα μέγιστορυθμό αναπνοής. Καθώς ο ρυθμός αναπνοής τουχρήστη αυξάνεται κατά τη διάρκεια δραστηριοτήτων
όπως το περπάτημα, η λειτουργία autoSATπροσαρμόζει τη λειτουργία της συσκευής ώστενα διατηρείται ένας καθορισμένος όγκος bolus.
UltraSense®
Αυτό το ευαίσθητο έναυσμα παλμικής δόσης ανιχνεύει
την αλλαγή πίεσης κατά τη διάρκεια της αναπνοής
σας και απελευθερώνει τον παλμό οξυγόνου ότανεισπνέετε.

Αναπνευστικός
ρυθμός
15
20
25
30
35
40

Ρύθμιση όγκου παλμού autoSat (mL)*
1
2
3
4
5
10,5
10,5
10,5
10,5
10,5
10,5

21,0
21,0
21,0
21,0
21,0
21,0

31,5
31,5
31,5
31,5
30,0
26,3

42,0
42,0
42,0
35,0
30,0
26,3

*Η ανοχή είναι +/-15%

388 - GR PN MN232-C4 L | Εγχειρίδιο χρήστη

52,5
52,5
42,0
35,0
30,0
26,3

autoDOSE®
Σε περίπτωση που προκύψει «συναγερμός
μη αναπνοής», αυτό το χαρακτηριστικό θα χορηγεί
παλμικό οξυγόνο σε ρυθμό 20 αναπνοών ανά λεπτό
στην τρέχουσα ρύθμιση ροής της μονάδας μέχρι ναεπιλυθεί η κατάσταση συναγερμού.

FreeStyle ComfortΤροφοδοτικά
Το FreeStyle Comfort μπορεί να τροφοδοτηθεί με τρεις
τρόπους:• Επαναφορτιζόμενο πακέτο μπαταρίας• Ισχύς AC• Ισχύς DCΠληροφορίες σχετικά με τη λειτουργία καθενός απόαυτά τα στοιχεία παρέχονται στις ακόλουθες ενότητες.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΧΡΗΣΙΜΟΠΟΙΕΊΤΕ
ΜΌΝΟ ΤΑ ΤΡΟΦΟΔΟΤΙΚΆ ΠΟΥ ΚΑΘΟΡΊΖΟΝΤΑΙ ΣΕ ΑΥΤΌ ΤΟ ΕΓΧΕΙΡΊΔΙΟ ΧΡΉΣΤΗ.
Η ΧΡΉΣΗ ΤΡΟΦΟΔΟΤΙΚΏΝ ΚΑΙ ΠΑΚΈΤΩΝ
ΜΠΑΤΑΡΊΑΣ ΆΛΛΩΝ ΑΠΌ CAIRE ΜΠΟΡΕΊΝΑ ΠΡΟΚΑΛΈΣΕΙ ΖΗΜΙΆ ΣΤΗ ΜΟΝΆΔΑ ΚΑΙΑΚΑΤΆΛΛΗΛΗ ΛΕΙΤΟΥΡΓΊΑ.

ΑΦΑΙΡΕΣΗ ΤΗΣ ΜΠΑΤΑΡΙΑΣ

Πιέστε το κουμπί απελευθέρωσης της μπαταρίας
προς τα κάτω μακριά από το FreeStyle Comfort.
Ενώ ΚΡΑΤΑΤΕ αυτό το κουμπί, ολισθήστε τηνμπαταρία προς τα έξω, τραβώντας την μακριάαπό τον συμπυκνωτή.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: Ο ΣΥΜΠΥΚΝΩΤΉΣ
ΟΞΥΓΌΝΟΥ ΠΡΈΠΕΙ ΝΑ ΑΠΕΝΕΡΓΟΠΟΙΗΘΕΙΚΑΙ ΝΑ ΑΠΟΣΥΝΔΕΘΕΊ ΑΠΌ ΤΗΝΕΞΩΤΕΡΙΚΉ ΤΡΟΦΟΔΟΣΊΑ ΡΕΎΜΑΤΟΣ ΠΡΙΝΤΗΝ ΑΦΑΊΡΕΣΗ ΤΟΥ ΠΑΚΈΤΟΥ ΜΠΑΤΑΡΊΑΣ.

ΠΑΚΕΤΟ ΜΠΑΤΑΡΙΑΣ

Το FreeStyle Comfort μπορεί να λειτουργήσει με
το επαναφορτιζόμενο πακέτο μπαταρίας, το οποίο
παρέχεται με τη συσκευή. Δεν υπάρχουν λειτουργικοί
περιορισμοί του συμπυκνωτή όταν λειτουργεί απότο πακέτο μπαταρίας. Όλοι οι ρυθμοί ροής είναι
διαθέσιμοι με ισχύ από την μπαταρία.

ΕΛΕΓΧΟΣ ΤΗΣ ΦΟΡΤΙΣΗΣ ΤΟΥ ΠΑΚΕΤΟΥΜΠΑΤΑΡΙΑΣΕΓΚΑΤΑΣΤΑΣΗ ΤΗΣ ΜΠΑΤΑΡΙΑΣ

Ευθυγραμμίστε την μπαταρία με το κάτω περίβλημααπό το κέντρο του FreeStyle Comfort. Ωθήστε τηνμπαταρία προς τα μέσα μέχρι να ακούσετε ένα κλικ γιανα επιβεβαιώσετε ότι έχει κλειδώσει στη θέση της.

Για να ελέγξετε τη φόρτιση της μπαταρίας ότανείναι εγκατεστημένη στο FreeStyle Comfort,συμβουλευτείτε το εικονίδιο μπαταρίας και τονεκτιμώμενο υπολειπόμενο χρόνο στην οθόνη.
80-100%60-80%

40-60%

20-40%

20%

PN MN232-C4 L | Εγχειρίδιο χρήστη GR - 389

FreeStyle ComfortΣημείωση: Εάν το τροφοδοτικό είναι συνδεδεμένο, θαεμφανίζεται μια ένδειξη βύσματος αντί για το υπολειπόμενο ποσοστό φόρτισης.
Για να ελέγξετε τη φόρτιση της μπαταρίας όταν δεν είναι
εγκατεστημένη στο FreeStyle Comfort, ακουμπίστε τηνεπίπεδα και πατήστε το κουμπί «TEST». Οι ενδεικτικές
λυχνίες μετρητή μπαταρίας (25-100%) θα ανάψουν σταδεξιά του κουμπιού TEST υποδεικνύοντας το επίπεδο τηςφόρτισης του πακέτου μπαταρίας.

ΑΡΧΙΚΗ ΦΟΡΤΙΣΗΗ νέα μπαταρία που παρέχεται με το FreeStyle
Comfort είναι μόνο μερικώς φορτισμένη όταναποστέλλεται από το εργοστάσιο. Πριν τη χρήση γιαπρώτη φορά, πρέπει να φορτίσετε πλήρως το πακέτο
μπαταρίας χρησιμοποιώντας είτε ισχύ AC είτε DC.

ΓΕΝΙΚΕΣ ΠΛΗΡΟΦΟΡΙΕΣ ΦΟΡΤΙΣΗΣ

Το FreeStyle Comfort θα επαναφορτίζει το πακέτο
μπαταρίας κάθε φορά που είναι εγκατεστημένο στον
συμπυκνωτή και ο συμπυκνωτής είναι συνδεδεμένος
σε μια εξωτερική πηγή ισχύος AC ή DC (εκτός εάνβρίσκεται σε αεροσκάφος).
Η μπαταρία είναι πλήρως φορτισμένη όταν εμφανίζει
την ένδειξη 100%.
Εναλλακτικά, το πακέτο μπαταρίας μπορεί ναφορτιστεί χρησιμοποιώντας τον προαιρετικό
επιτραπέζιο φορτιστή.

390 - GR PN MN232-C4 L | Εγχειρίδιο χρήστη

ΔΙΑΡΚΕΙΑ ΛΕΙΤΟΥΡΓΙΑΣ ΜΕ ΜΠΑΤΑΡΙΑ

Όταν αποσυνδεθεί η εξωτερική ισχύς AC ή DC,
ο συμπυκνωτής θα αλλάξει αυτόματα σε χρήση
ισχύος από την μπαταρία εάν έχει εγκατασταθεί έναφορτισμένο πακέτο μπαταρίας. Ο πίνακας παρακάτω
δείχνει την τυπική διάρκεια για ένα καινούριο πακέτο
μπαταρίας. Η διάρκεια του πακέτου μπαταρίας θαπαραμείνει σταθερή ανεξάρτητα από τον αναπνευστικό
ρυθμό σας.
Ρύθμισηπαλμού

Διάρκεια μονής
μπαταρίας
(8 στοιχείων)Διάρκεια διπλής
μπαταρίας
(16 στοιχείων)1

έως 8 ώρεςέως 16 ώρες2

έως 4 ώρεςέως 8 ώρες3

έως 3 ώρεςέως 6 ώρες4

έως 2,25 ώρεςέως 4,5 ώρες5

έως 2 ώρεςέως 4 ώρεςΤΥΠΙΚΟΣ ΧΡΟΝΟΣ ΕΠΑΝΑΦΟΡΤΙΣΗΣΜΠΑΤΑΡΙΑΣΟ τυπικός χρόνος για την πλήρη επαναφόρτιση τουπακέτου μπαταρίας από μια πλήρως αποφορτισμένη
μπαταρία είναι 3,5 ώρες για μια μονή μπαταρία και6,0 ώρες για μια διπλή μπαταρία.
Σημείωση: Η μπαταρία μπορεί να μην ξεκινήσει ναφορτίζεται αρχικά εάν το πακέτο μπαταρίας έχειεξαντληθεί πλήρως.

ΩΦΕΛΙΜΗ ΔΙΑΡΚΕΙΑ ΖΩΗΣ ΜΠΑΤΑΡΙΑΣ
Οι μπαταρίες του FreeStyle Comfort είναι
σχεδιασμένες για να διαρκούν για 300 κύκλους
φόρτισης/αποφόρτισης.

FreeStyle ComfortΤροφοδοτικό AC και DC
Το FreeStyle Comfort διατίθεται τυπικά μετροφοδοτικά AC και DC.
Το τροφοδοτικό AC περιλαμβάνει τα ακόλουθα:	α. Μονάδα τροφοδοτικού AC που λειτουργεί
μεταξύ 100-240 volts, 50-60Hz.
	β. Καλώδιο ρεύματος AC για τη σύνδεση στοτροφοδοτικό και στην αντίστοιχη παροχή AC.

Το τροφοδοτικό DC αποτελείται από ένα καλώδιο πουσυνδέεται απευθείας στη μονάδα και στην παροχή DC.

ΙΣΧΥΣ ACΗ ισχύς AC προορίζεται για χρήση στο σπίτι ήοπουδήποτε είναι διαθέσιμη ισχύς παροχής AC.
Το FreeStyle Comfort είναι πλήρως λειτουργικό
με ισχύ AC. Το FreeStyle Comfort επαναφορτίζει
επίσης το πακέτο μπαταρίας οποιαδήποτε στιγμή
είναι διαθέσιμη ισχύς AC (εκτός εάν βρίσκεται σεαεροσκάφος). Το πακέτο μπαταρίας θα επαναφορτιστεί
είτε το FreeStyle Comfort είναι ενεργοποιημένο είτεαπενεργοποιημένο.

2. Συνδέστε το καλώδιο τροφοδοσίας AC στοτροφοδοτικό AC.

ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΣΥΝΔΈΕΤΕ ΤΗΝΕΞΩΤΕΡΙΚΉ ΙΣΧΎ ΧΩΡΊΣ ΝΑ ΥΠΆΡΧΕΙ
ΕΓΚΑΤΕΣΤΗΜΈΝΟ ΈΝΑ ΠΑΚΈΤΟ
ΜΠΑΤΑΡΊΑΣ ΣΤΟΝ ΣΥΜΠΥΚΝΩΤΉ
ΟΞΥΓΌΝΟΥ.
Για τη σύνδεση του FreeStyle Comfort στην ισχύ AC:
1. Συνδέστε το τροφοδοτικό AC στην είσοδο
συνδέσμου ισχύος στην πρόσοψη, στην κάτω δεξιά
πλευρά του FreeStyle Comfort. Βεβαιωθείτε ότιωθήσατε προς τα μέσα έτσι ώστε η σύνδεση ναείναι ασφαλής.

3. Συνδέστε το αντίθετο άκρο του καλωδίου
τροφοδοσίας AC στην επιτοίχια πρίζα ή άλληκατάλληλη παροχή ρεύματος.

PN MN232-C4 L | Εγχειρίδιο χρήστη GR - 391

FreeStyle Comfort
4. Ενεργοποιήστε το FreeStyle Comfort πατώνταςτο κουμπί λειτουργίας και επιλέξτε τησυνταγογραφημένη ρύθμιση ροής χρησιμοποιώντας
τα κουμπιά αύξησης (+) ή μείωσης (-) ροής.

Σημείωση: Εάν το πακέτο μπαταρίας χρειάζεται επαναφόρτιση, το εικονίδιο μπαταρίας στην οθόνη θα έχει μίαμόνο πορτοκαλί ράβδο.

ΙΣΧΥΣ DCΗ ισχύς DC προορίζεται για χρήση στις πρίζες
παρελκομένων αυτοκινήτων, σκαφών ή άλλων
μηχανοκίνητων οχημάτων. Το FreeStyle Comfortείναι πλήρως λειτουργικό για χρήση με ισχύ DC.
Το FreeStyle Comfort επαναφορτίζει επίσης τοπακέτο μπαταρίας εάν χρειάζεται, οποιαδήποτε
στιγμή είναι διαθέσιμη ισχύς DC. Το πακέτο
μπαταρίας θα επαναφορτιστεί είτε η μονάδα είναι
απενεργοποιημένη, είτε είναι ενεργοποιημένη και σεχρήση.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΣΥΝΔΕΕΤΕ ΤΗΝΕΞΩΤΕΡΙΚΗ ΙΣΧΥ ΧΩΡΙΣ ΝΑ ΥΠΑΡΧΕΙ
ΕΓΚΑΤΕΣΤΗΜΕΝΗ ΣΥΣΤΟΙΧΙΑ ΜΠΑΤΑΡΙΩΝΣΤΟΝ ΣΥΜΠΥΚΝΩΤΗ ΟΞΥΓΟΝΟΥ.
1. Ενεργοποιήστε το μηχανοκίνητο όχημά σας.
2. Συνδέστε το τροφοδοτικό DC στην υποδοχή θύρας
τροφοδοσίας στην πρόσοψη, στην κάτω δεξιά
πλευρά του FreeStyle Comfort. Βεβαιωθείτε ότιωθήσατε προς τα μέσα έτσι ώστε η σύνδεση ναείναι ασφαλής.

3. Συνδέστε το αντίθετο άκρο του καλωδίου
τροφοδοσίας DC στην πρίζα του αυτοκινήτουή άλλη παρόμοια παροχή αξεσουάρ DC.
392 - GR PN MN232-C4 L | Εγχειρίδιο χρήστη

4. Ενεργοποιήστε το FreeStyle Comfort πατώνταςτο κουμπί λειτουργίας και επιλέξτε τησυνταγογραφημένη ρύθμιση ροής χρησιμοποιώντας
τα κουμπιά αύξησης (+) ή μείωσης (-) ροής.
ΠΡΟΣΟΧΗ: Για την πρόληψη πιθανής
πυρκαγιάς ή υπερθέρμανσης, βεβαιωθείτεότι η πρίζα των 12V DC είναι ελεύθερηαπό υπολείμματα και δεν καλύπτεταικαι ότι το βύσμα του προσαρμογέα τουτροφοδοτικού DC εφαρμόζει σωστά και
παραμένει σταθερά στη θέση του.
ΠΡΟΣΟΧΗ: Οι μεταλλικές επιφάνειες τουτροφοδοτικού DC μπορεί να γίνουν ΠΟΛΥ ΚΑΥΤΕΣ
κατά τη χρήση τους λόγω του υψηλού ρεύματος.
Αφήστε να περάσει επαρκής χρόνος για να κρυώσουν
οι επιφάνειες πριν τις αγγίξετε.
ΠΡΟΣΟΧΗ: Μην χρησιμοποιείτε το τροφοδοτικό DCμε επεκτάσεις, διανομείς ή προσαρμογείς. Η χρήση μηεγκεκριμένων παρελκομένων μπορεί να οδηγήσει σευπερθέρμανση ή καταστροφή τους.
ΠΡΟΣΟΧΗ: Η σύνδεση της συσκευής σε πρίζα των
12V DC μπορεί να εξαντλήσει την μπαταρία τουοχήματος, αν ο κινητήρας δεν λειτουργεί, ακόμη καιαν η συσκευή είναι απενεργοποιημένη.
ΠΡΟΣΟΧΗ: Μην προβαίνετε σε ρευματοδότηση τουοχήματος από άλλο όχημα, ενώ είναι συνδεδεμένοτο τροφοδοτικό DC.

FreeStyle Comfort
Καθαρισμός, φροντίδα καισυντήρηση ρουτίνας
ΣΥΝΤΗΡΗΣΗ ΡΟΥΤΙΝΑΣΠροληπτική συντήρηση
Η συσκευή δεν χρειάζεται συντήρηση. Τα φίλτρα στοεσωτερικό της συσκευής προορίζονται να έχουν διάρκεια
για τη διάρκεια ζωής της συσκευής. Εάν η συσκευή δενλειτουργεί σωστά, η μονάδα πρέπει να επιστραφεί στονπρομηθευτή για επισκευή. Η συντήρηση των εσωτερικών
εξαρτημάτων στο εσωτερικό του περιβλήματος τουFreeStyle Comfort πρέπει να πραγματοποιείται από τονπάροχο οξυγόνου.
Ο χειριστής της μονάδας είναι υπεύθυνος για τηνακόλουθη διαδικασία συντήρησης.
• Εβδομαδιαία – Καθαρισμός των φίλτρωνεισαγωγής αέραΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗΝ ΕΠΙΧΕΙΡΉΣΕΤΕΝΑ ΑΝΟΊΞΕΤΕ Ή ΝΑ ΑΦΑΙΡΈΣΕΤΕ ΤΑΠΛΑΣΤΙΚΆ ΚΑΛΎΜΜΑΤΑ ΤΟΥ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ. Η ΜΟΝΆΔΑ ΠΡΈΠΕΙ ΝΑΑΝΟΊΓΕΤΑΙ ΜΌΝΟ ΑΠΌ ΈΝΑΝ ΕΙΔΙΚΕΥΜΈΝΟ ΤΕΧΝΙΚΌ ΣΈΡΒΙΣ.

ΦΙΛΤΡΑ ΕΙΣΑΓΩΓΗΣ ΑΕΡΑΟ αέρας εισέρχεται στο FreeStyle Comfort μέσωτων φίλτρων εισαγωγής αέρα που βρίσκονται στηνπρόσοψη, στο κάτω μέρος του συμπυκνωτή οξυγόνου.
Αυτά τα φίλτρα εμποδίζουν τη σκόνη και άλλα μεγάλα
σωματίδια στον αέρα να εισέλθουν στη μονάδα.
Τουλάχιστον μία φορά την εβδομάδα, τα φίλτρα
εισαγωγής αέρα πρέπει να καθαρίζονται. Ο πάροχος
οξυγόνου σας μπορεί να σας συμβουλεύσει να τακαθαρίζετε συχνότερα, ανάλογα με τις συνθήκες
λειτουργίας. Ακολουθήστε αυτά τα βήματα για τονσωστό καθαρισμό των φίλτρων εισαγωγής αέρα.
1. Αφαιρέστε τα δύο φίλτρα από την πρόσοψη τουFreeStyle Comfort ωθώντας προς τα πάνω τηναντίστοιχη γλωττίδα και τραβώντας προς τα έξω.

ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΛΕΙΤΟΥΡΓΕΊΤΕ
ΤΟΝ ΣΥΜΠΥΚΝΩΤΉ ΟΞΥΓΌΝΟΥ ΧΩΡΊΣΝΑ ΕΊΝΑΙ ΤΟΠΟΘΕΤΗΜΈΝΟ ΤΟ ΦΊΛΤΡΟ
ΕΙΣΑΓΩΓΉΣ ΑΈΡΑ.

ΦΡΟΝΤΙΔΑ ΤΟΥ ΠΑΚΕΤΟΥ ΜΠΑΤΑΡΙΑΣ

Μην αφήνετε το πακέτο μπαταρίας να εκτεθεί σε υγρά.
Μην επιχειρήσετε να αποσυναρμολογήσετε ή ναανοίξετε το πακέτο μπαταρίας.
Μην εκθέτετε την μπαταρία σε γυμνές φλόγες.
Μην αφήνετε το πακέτο μπαταρίας να πέσει.
Φυλάσσετε το πακέτο μπαταρίας μακριά από παιδιά.
Εάν το πακέτο μπαταρίας φαίνεται να έχει υποστεί
ζημιά, επικοινωνήστε με τον πάροχο οξυγόνου σαςπριν τη χρήση.
Μην τοποθετείτε την μπαταρία μέσα ή κοντά σεφούρνο μικροκυμάτων ή άλλες μαγειρικές συσκευές.

ΑΠΟΘΗΚΕΥΣΗ ΤΟΥ ΠΑΚΕΤΟΥΜΠΑΤΑΡΙΑΣΗ μπαταρία του συμπυκνωτή οξυγόνου δενχρειάζεται να έχει αποφορτιστεί πλήρως πριν από τηνεπαναφόρτιση. Συνιστάται η φόρτιση της μπαταρίας
του συμπυκνωτή οξυγόνου μετά από κάθε χρήση.
Μην αφήνετε την μπαταρία σε μέρος εκτεθειμένο
σε υπερβολική θερμότητα, όπως σε άμεση ηλιακή
ακτινοβολία ή μέσα σε αυτοκίνητο. Αυτό θα μπορούσε
να προκαλέσει υποβάθμιση της απόδοσης τηςμπαταρίας και της διάρκειας ζωής της. Μπορεί επίσης
να προκαλέσει έκλυση θερμότητας ή καπνού απότην μπαταρία, ανάφλεξη ή έκρηξη της μπαταρίας.
Το πακέτο μπαταρίας πρέπει να αποθηκεύεται
σε σταθερό περιβάλλον που χαρακτηρίζεται απόχαμηλή υγρασία (ΣΥ υγρασία χαμηλότερη από70%) και απουσία διαβρωτικών αερίων. Για τηνπρόληψη της σκουριάς, αποφεύγετε συνθήκες πουμπορεί να δημιουργήσουν συμπύκνωση υδρατμών,όπως γρήγορες διακυμάνσεις της θερμοκρασίας τουπεριβάλλοντος.
Για αποθήκευση άνω του ενός μηνός, συνιστάται
αποφορτισμένη ή μερικώς αποφορτισμένη κατάστασηφόρτισης, μεταξύ 50% και 75% (δύο έως τρία LED).
Όταν είναι φορτισμένη, αφαιρέστε την μπαταρία απότον συμπυκνωτή οξυγόνου.

2. Πλύνετε και τα δύο φίλτρα σε ζεστό διάλυμα
σαπουνιού και νερού.
3. Ξεπλύνετε τα φίλτρα τελείως και αφαιρέστε τοπερίσσιο νερό με ένα μαλακό πανί που δεν αφήνει
χνούδι. Βεβαιωθείτε ότι τα φίλτρα είναι τελείως
στεγνά πριν τα επανατοποθετήσετε.
4. Επανατοποθετήστε τα καθαρά, στεγνά φίλτρα.
PN MN232-C4 L | Εγχειρίδιο χρήστη GR - 393

FreeStyle ComfortΚΑΘΑΡΙΣΜΟΣΑΠΕΝΕΡΓΟΠΟΙΗΣΤΕ τη μονάδα και αποσυνδέστε
την από την τροφοδοσία AC ή DC πριν απόοποιαδήποτε διαδικασία καθαρισμού. ΜΗΝ ψεκάζετε
απευθείας στο εξωτερικό περίβλημα. Χρησιμοποιήστεένα υγραμένο (όχι μουσκεμένο) πανί ή σφουγγάρι.
Ψεκάστε το πανί ή το σφουγγάρι με ένα διάλυμα ήπιου
καθαριστικού οικιακής χρήσης για να καθαρίσετε τοπερίβλημα και τα τροφοδοτικά. Προχωρήστε σύμφωναμε τις οδηγίες του κατασκευαστή του καθαριστικού.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΑΠΕΝΕΡΓΟΠΟΙΗΣΤΕ ΤΗΜΟΝΑΔΑ ΚΑΙ ΑΠΟΣΥΝΔΕΣΤΕ ΤΟ ΚΑΛΩΔΙΟ
ΡΕΥΜΑΤΟΣ ΑΠΟ ΤΗΝ ΗΛΕΚΤΡΙΚΗ ΠΑΡΟΧΗ
ΠΡΙΝ ΤΟΝ ΚΑΘΑΡΙΣΜΟ ΤΗΣ ΜΟΝΑΔΑΣ
ΠΡΟΚΕΙΜΕΝΟΥ ΝΑ ΑΠΟΦΥΓΕΤΕΤΥΧΟΝ ΗΛΕΚΤΡΟΠΛΗΞΙΑ ΚΑΙ ΚΙΝΔΥΝΟΕΓΚΑΥΜΑΤΩΝ. ΜΟΝΟ Ο ΠΑΡΟΧΟΣ ΤΟΥΕΞΟΠΛΙΣΜΟΥ ΣΑΣ Ή ΕΝΑΣ ΕΙΔΙΚΕΥΜΕΝΟΣΤΕΧΝΙΚΟΣ ΣΕΡΒΙΣ ΕΠΙΤΡΕΠΕΤΑΙ ΝΑΑΦΑΙΡΕΣΕΙ ΤΑ ΚΑΛΥΜΜΑΤΑ Ή ΝΑΣΥΝΤΗΡΗΣΕΙ ΤΗ ΜΟΝΑΔΑ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΠΡΕΠΕΙ ΝΑ ΛΑΜΒΑΝΕΤΑΙΜΕΡΙΜΝΑ ΩΣΤΕ ΝΑ ΜΗ ΒΡΑΧΕΙ Ο ΣΥΜΠΥΚΝΩΤΗΣΟΞΥΓΟΝΟΥ ΚΑΙ ΤΟ ΠΑΚΕΤΟ ΜΠΑΤΑΡΙΑΣ Ή ΝΑΜΗ ΔΙΕΙΣΔΥΣΟΥΝ ΥΓΡΑ ΣΤΗ ΜΟΝΑΔΑ. ΚΑΤΙΤΕΤΟΙΟ ΜΠΟΡΕΙ ΝΑ ΠΡΟΚΑΛΕΣΕΙ ΔΥΣΛΕΙΤΟΥΡΓΙΑΉ ΑΠΕΝΕΡΓΟΠΟΙΗΣΗ, ΚΑΘΩΣ ΚΑΙ ΑΥΞΗΜΕΝΟΚΙΝΔΥΝΟ ΗΛΕΚΤΡΟΠΛΗΞΙΑΣ Ή ΕΓΚΑΥΜΑΤΩΝ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗ ΧΡΗΣΙΜΟΠΟΙΕΙΤΕ ΥΓΡΟΑΠΕΥΘΕΙΑΣ ΕΠΑΝΩ ΣΤΗ ΜΟΝΑΔΑ. ΕΝΑΣ ΚΑΤΑΛΟΓΟΣΑΝΕΠΙΘΥΜΗΤΩΝ ΧΗΜΙΚΩΝ ΟΥΣΙΩΝ ΠΕΡΙΛΑΜΒΑΝΕΙ,ΑΛΛΑ ΟΧΙ ΠΕΡΙΟΡΙΣΤΙΚΑ, ΤΑ ΑΚΟΛΟΥΘΑ: ΑΛΚΟΟΛΗΚΑΙ ΠΡΟΪΟΝΤΑ ΜΕ ΒΑΣΗ ΑΛΚΟΟΛΗ, ΣΥΜΠΥΚΝΩΜΕΝΑ
ΠΡΟΪΟΝΤΑ ΜΕ ΒΑΣΗ ΧΛΩΡΙΟ (ΧΛΩΡΙΟΥΧΟ
ΑΙΘΥΛΕΝΙΟ) ΚΑΙ ΠΡΟΪΟΝΤΑ ΜΕ ΒΑΣΗ ΛΑΔΙ. ΑΥΤΑ ΔΕΝΠΡΕΠΕΙ ΝΑ ΧΡΗΣΙΜΟΠΟΙΟΥΝΤΑΙ ΓΙΑ ΤΟΝ ΚΑΘΑΡΙΣΜΟ
ΤΟΥ ΠΛΑΣΤΙΚΟΥ ΠΕΡΙΒΛΗΜΑΤΟΣ ΤΟΥ ΣΥΜΠΥΚΝΩΤΗ,ΚΑΘΩΣ ΜΠΟΡΕΙ ΝΑ ΠΡΟΚΑΛΕΣΟΥΝ ΖΗΜΙΑ ΣΤΗΜΟΝΑΔΑ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΚΑΘΑΡΙΣΤΕ ΤΟ ΠΕΡΙΒΛΗΜΑ,ΤΟΝ ΠΙΝΑΚΑ ΕΛΕΓΧΟΥ ΚΑΙ ΤΟ ΚΑΛΩΔΙΟ ΡΕΥΜΑΤΟΣΜΟΝΟ ΜΕ ΕΝΑ ΗΠΙΟ ΟΙΚΙΑΚΟ ΚΑΘΑΡΙΣΤΙΚΟ ΠΟΥΑΠΛΩΝΕΤΕ ΜΕ ΥΓΡΑΜΕΝΟ (ΟΧΙ ΜΟΥΣΚΕΜΕΝΟ)ΠΑΝΙ Ή ΣΦΟΥΓΓΑΡΙ, ΚΑΙ ΣΤΗ ΣΥΝΕΧΕΙΑ ΣΤΕΓΝΩΣΤΕΣΚΟΥΠΙΖΟΝΤΑΣ ΟΛΕΣ ΤΙΣ ΕΠΙΦΑΝΕΙΕΣ. ΜΗΝΕΠΙΤΡΕΠΕΤΕ ΤΗ ΔΙΕΙΣΔΥΣΗ ΟΠΟΙΟΥΔΗΠΟΤΕΥΓΡΟΥ ΣΤΟ ΕΣΩΤΕΡΙΚΟ ΤΟΥ ΣΥΜΠΥΚΝΩΤΗ. ΔΩΣΤΕΙΔΙΑΙΤΕΡΗ ΠΡΟΣΟΧΗ ΣΤΗΝ ΕΞΟΔΟ ΟΞΥΓΟΝΟΥ ΓΙΑΤΗ ΣΥΝΔΕΣΗ ΤΗΣ ΚΑΝΟΥΛΑΣ ΠΡΟΚΕΙΜΕΝΟΥ ΝΑΔΙΑΣΦΑΛΙΣΕΤΕ ΟΤΙ ΠΑΡΑΜΕΝΕΙ ΑΠΑΛΛΑΓΜΕΝΗ ΑΠΟΣΚΟΝΗ, ΝΕΡΟ ΚΑΙ ΣΩΜΑΤΙΔΙΑ.

394 - GR PN MN232-C4 L | Εγχειρίδιο χρήστη

ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΧΡΗΣΙΜΟΠΟΙΗΣΤΕ ΕΝΑΥΓΡΑΜΕΝΟ (ΌΧΙ ΜΟΥΣΚΕΜΕΝΟ) ΠΑΝΙ Ή ΣΦΟΥΓΓΑΡΙ
ΓΙΑ ΤΟΝ ΚΑΘΑΡΙΣΜΟ ΤΟΥ ΠΑΚΕΤΟΥ ΜΠΑΤΑΡΙΑΣ.
ΨΕΚΑΣΤΕ ΤΟ ΠΑΝΙ Ή ΤΟ ΣΦΟΥΓΓΑΡΙ ΜΕ ΕΝΑ ΗΠΙΟΟΙΚΙΑΚΟ ΚΑΘΑΡΙΣΤΙΚΟ ΓΙΑ ΝΑ ΚΑΘΑΡΙΣΕΤΕ ΤΟΠΕΡΙΒΛΗΜΑ ΤΟΥ ΠΑΚΕΤΟΥ ΜΠΑΤΑΡΙΑΣ.
ΠΡΟΕΙΔΟΠΟΙΗΣΗ: ΜΗΝ ΚΑΘΑΡΊΖΕΤΕ ΜΕΑΥΤΟΜΑΤΟΠΟΙΗΜΈΝΟ ΣΎΣΤΗΜΑ ΠΛΎΣΗΣ.

ΑΝΤΙΚΑΤΑΣΤΑΣΗ ΤΗΣ ΚΑΝΟΥΛΑΣ

Πάντα να ακολουθείτε τις οδηγίες του κατασκευαστήτης κάνουλας για σωστή χρήση. Αντικαθιστάτε τηρινική κάνουλα ή τη σωλήνωση οξυγόνου κάθε3 έως 6 μήνες (με καθημερινή χρήση, 3 μήνες) όπωςσυνιστάται από τον κατασκευαστή της κάνουλας ήτον πάροχο οξυγόνου σας. Ο γιατρός ή ο πάροχος
οξυγόνου σας θα σας παράσχουν οδηγίες καθαρισμού
και αντικατάστασης.
Επιπλέον αναλώσιμα για αντικατάσταση είναι
διαθέσιμα μέσω του παρόχου οξυγόνου σας.

FreeStyle Comfort
Καταστάσεις συναγερμού
Αρ. προτεραιότητας

Περιγραφή
συναγερμού

1Δυσλειτουργία
του συστήματος - Υψηλή
θερμοκρασία
μπαταρίας

Εμφανιζόμενη ένδειξηΗχητικόσήμαΠερισσότερες λεπτομέρειες / Προτεινόμενη ενέργεια

1 μπιπκάθε
20 δευτερόλεπταΗ μονάδα θα παράγει συναγερμό ότανη θερμοκρασία της μπαταρίας πλησιάζει τοόριο διακοπής λειτουργίας ασφαλείας.
Εάν επιτευχθεί το όριο απενεργοποίησης,η μονάδα θα απενεργοποιηθεί. Εάν είναι
συνδεδεμένη εξωτερική τροφοδοσία, θα εμφανιστεί η οθόνη συναγερμού και ο ηχητικός
συναγερμός θα συνεχιστεί για 2 λεπτά.
Πραγματοποιήστε δοκιμή σε θερμοκρασία
δωματίου. Εάν το πρόβλημα επιμένει, επικοινωνήστε με τον πάροχο.

2Δυσλειτουργία
τουσυστήματος Υψηλή
θερμοκρασία

1 μπιπκάθε
20 δευτερόλεπταΗ μονάδα θα παράγει συναγερμό όταν ηεσωτερική θερμοκρασία πλησιάζει το όριοαπενεργοποίησης.
Εάν επιτευχθεί το όριο απενεργοποίησης,η μονάδα θα απενεργοποιηθεί και θα εμφανιστεί η οθόνη συναγερμού και ο ηχητικός
συναγερμός θα συνεχιστεί για 2 λεπτά.
Διασφαλίστε ότι δεν υπάρχουν εμφράξεις
σε κανένα άνοιγμα αερισμού της μονάδας.
Εάν το πρόβλημα επιμένει, επικοινωνήστε
με τον πάροχο.

3Δυσλειτουργία
συστήματοςΥψηλή πίεση
(υψηλή πίεση
συστήματοςή βλάβησυμπιεστή)1 μπιπκάθε
20 δευτερόλεπτα

4Δυσλειτουργία
του συστήματος - Ρυθμός
ροής (βλάβη
βαλβίδαςOCD, έμφραξη
διαδρομής
αερίου)1 μπιπκάθε
30 δευτερόλεπταΕάν το πρόβλημα επιμένει, επικοινωνήστε
με τον πάροχο.

5Χαμηλή
μπαταρία απενεργοποίηση

1 μπιπκάθε 30δευτερόλεπταΟ συναγερμός παράγεται μέχρι την εξάντληση της μπαταρίας και την απενεργοποίηση
της μονάδας.

Εάν επιτευχθεί το όριο απενεργοποίησης,η μονάδα θα απενεργοποιηθεί και θαεμφανιστεί η οθόνη συναγερμού και οηχητικός συναγερμός θα συνεχιστεί για2 λεπτά.
Εάν το πρόβλημα επιμένει, επικοινωνήστε
με τον πάροχο.

Φορτίστε την μπαταρία.

PN MN232-C4 L | Εγχειρίδιο χρήστη GR - 395

FreeStyle ComfortΑρ. προτεραιότητας

Περιγραφή
συναγερμού6Δεν ανιχνεύτηκε αναπνοή

Εμφανιζόμενη ένδειξηΗχητικόσήμαΠερισσότερες λεπτομέρειες

1 μπιπκάθε
10 λεπτά

Δεν ανιχνεύτηκε αναπνοή για 15 δευτερόλεπτα.
Ενεργοποιείται μετά την πρώτη αναπνοή ήαφού απενεργοποιηθεί η οθόνη προθέρμανσης, όποιο από τα δύο συμβεί πρώτο.
Ελέγξτε την κάνουλα για συστροφές ήδιαρροές. Αυξήστε τη ρύθμιση ευαισθησίας.
Εάν το πρόβλημα επιμένει, επικοινωνήστε μετον πάροχο.

7ΠροθέρμανσηκανέναΗ μονάδα θα ξεκινήσει με αυτόματη χορήγηση μέχρι να ανιχνευτεί αναπνοή ή να λήξειη προθέρμανση, όποιο από τα δύο συμβεί
πρώτο.

8Χαμηλή
συγκέντρωση

1 μπιπκάθε
10 λεπτά,για έως
30 λεπτάΕάν το πρόβλημα επιμένει, επικοινωνήστε μετον πάροχο.

9Προειδοποίηση χαμηλής
μπαταρίας

1 μπιπ,χωρίςεπανάληψηΦορτίστε την μπαταρία.

10Υπέρβαση τουαναπνευστικού ρυθμού

1 μπιπ,χωρίςεπανάληψηΟ αναπνευστικός ρυθμός υπερβαίνει τα40 bpm (αναπνοές ανά λεπτό).

11Δυσλειτουργία
του συστήματος - Βλάβη
ανεμιστήρα

κανέναΕάν το πρόβλημα επιμένει, επικοινωνήστε
με τον πάροχο.

12Δυσλειτουργία
του συστήματος - Χαμηλή
πίεσηκανέναΕάν το πρόβλημα επιμένει, επικοινωνήστε
με τον πάροχο.

13Δυσλειτουργία
του συστήματος - Χαμηλή
θερμοκρασία

κανέναΑφήστε τη μονάδα να προθερμανθεί σεθερμοκρασία δωματίου. Εάν το πρόβλημαεπιμένει, επικοινωνήστε με τον πάροχο.

396 - GR PN MN232-C4 L | Εγχειρίδιο χρήστηΕάν η δόση ενεργοποιείται εσφαλμένα,μειώστε τη ρύθμιση ευαισθησίας. Εάν τοπρόβλημα επιμένει, επικοινωνήστε με τονπάροχο.

FreeStyle ComfortΑρ. προτεραιότητας

Περιγραφή
συναγερμού

Πληροφοριακό

ΕκκίνησηΕμφανιζόμενη ένδειξηΗχητικόσήμαΠερισσότερες λεπτομέρειες

1 πληροφοριακό
μπιπ

Η οθόνη εμφανίζεται κατά την ενεργοποίηση
και θα πρέπει να ακούσετε ένα μπιπ για ναεπιβεβαιωθεί ότι ο βομβητής λειτουργεί.
Η οθόνη εκκίνησης εμφανίζει την τρέχουσα
αναθεώρηση υλικολογισμικού στην κάτωδεξιά γωνία.

Πληροφοριακό

Οθόνη κανονικής λειτουργίας σε τρόπο
autoSAT(χωρίς
συνδεδεμένη
τροφοδοσία
DC)κανέναΠληροφοριακόΟθόνηκανονικής
λειτουργίας
με το autoSAT
απενεργοποιημένο (χωρίς
συνδεδεμένη
τροφοδοσία
DC)

Καμία

ΠληροφοριακόΗ ρύθμισηάλλαξε1 πληροφοριακόμπιπΠληροφοριακό

Σύνδεσηεξωτερικής
τροφοδοσίας
στην απενεργοποιημένη
κατάστασηκανέναΠληροφοριακό

Σύνδεσηεξωτερικής
τροφοδοσίαςστηνκατάσταση
λειτουργίας

κανέναΠληροφοριακό

Αποσύνδεση
εξωτερικής
τροφοδοσίας
στην απενεργοποιημένη
κατάσταση1 πληροφοριακό
μπιπΕμφάνιση της οθόνης CAIRE για 2 δευτερόλεπτα, ακολουθούμενη από την κατάσταση
μπαταρίας και σύνδεσης για 5 δευτερόλεπταπριν απενεργοποιηθεί η οθόνη.

Εμφάνιση της οθόνης CAIRE για 2 δευτερόλεπτα και στη συνέχεια απενεργοποίηση τηςοθόνης.

PN MN232-C4 L | Εγχειρίδιο χρήστη GR - 397

FreeStyle ComfortΑρ. προτεραιότητας

Περιγραφή
συναγερμού

Εμφανιζόμενη ένδειξηΠληροφοριακό

Αποσύνδεση
εξωτερικής
τροφοδοσίας
στην κατάσταση λειτουργίας
(με παρουσία
μπαταρίας)1 πληροφοριακόμπιπΠληροφοριακό

Στην κατάσταση αναμονής όταν πατηθείτο κουμπί
Πάνω ή Κάτω,η οθόνηεμφανίζει τηνκατάσταση
μπαταρίας καιφόρτισηςκανέναΠληροφοριακό

Λειτουργία
πτήσης μετροφοδοσία
από μπαταρία

1 μακρύμπιπΠληροφοριακό

Λειτουργία
πτήσης
ενώ είναι
συνδεδεμένο

1 μακρύ
μπιπ

398 - GR PN MN232-C4 L | Εγχειρίδιο χρήστη

ΗχητικόσήμαΠερισσότερες λεπτομέρειεςΗ οθόνη θα εμφανιστεί για 5 δευτερόλεπτακαι στη συνέχεια θα απενεργοποιηθεί.

FreeStyle ComfortΟδηγίες πλοήγησης μενού
Μπορείτε να μπείτε στο μενού σε κατάσταση
αναμονής ή λειτουργίας. Σε κατάσταση λειτουργίας,εάν παρουσιαστεί κατάσταση συναγερμού, θα γίνει
αυτόματα έξοδος από το μενού. Εάν δεν πατηθεί κανένα
κουμπί για 20 δευτερόλεπτα, θα γίνει αυτόματα έξοδοςαπό το μενού. Σε κατάσταση αναμονής, μπορεί ναχρειαστεί να πατήσετε το κουμπί μενού δύο φορές γιανα το αφυπνίσετε, στη συνέχεια μπείτε στο μενού.
Υπάρχουν τέσσερις οθόνες μενού:1. Οθόνη Status (Κατάσταση): Εμφανίζει τον αριθμό
έκδοσης υλικολογισμικού, τον χρόνο λειτουργίας,τους κύκλους αποφόρτισης της μπαταρίας

Airplane Mode (Λειτουργία πτήσης)2. Airplane Mode (Λειτουργία πτήσης) (ON/OFF ΕΝΕΡΓΗ/ΑΝΕΝΕΡΓΗ)
3. Λειτουργία Sensitivity (Ευαισθησία): ρυθμιζόμενη
(1-5)
4. autoSAT (ON/OFF - Ενεργό/Ανενεργό)Sensitivty (Ευαισθησία)Παράδειγμα οθόνης Status (Κατάσταση)ΛΕΙΤΟΥΡΓΊΑ ΜΕΝΟΎ ΡΥΘΜΊΣΕΩΝ

autoSATΗ είσοδος στο μενού ρυθμίσεων γίνεται πατώντας
σε κατάστασητο κουμπί Μενού/Επιλογή
λειτουργίας ή αναμονής.
Για κύλιση στα μενού, πατήστε το.

Σελίδα Status (Κατάσταση)PN MN232-C4 L | Εγχειρίδιο χρήστη GR - 399

FreeStyle ComfortΣΕΛΊΔΑ STATUS (ΚΑΤΆΣΤΑΣΗ)Η σελίδα Status (Κατάσταση) του Freestyle Comfortδείχνει την αναθεώρηση υλικολογισμικού, τις ώρεςλειτουργίας και τον αριθμό των κύκλων μπαταρίας τηςσυνδεδεμένης μπαταρίας.

Για να ακυρώσετε την ενεργοποίηση της λειτουργίας
ενόσω το εικονίδιο CANCEL
πτήσης, πατήστε
(Άκυρο) είναι ορατό.

Επιβεβαιωμένη ενεργή λειτουργία πτήσης

Σελίδα Status (Κατάσταση)Οι ώρες λειτουργίας εμφανίζονται σε ολόκληρες ώρες.
Η ένδειξη Battery Discharges (Αποφορτίσεις
μπαταρίας) είναι ο αριθμός των κύκλων αποφόρτισης
και επαναφόρτισης της συνδεδεμένης μπαταρίας.

AIRPLANE MODE (ΛΕΙΤΟΥΡΓΊΑ ΠΤΉΣΗΣ)Το FreeStyle Comfort αποστέλλεται με τη λειτουργία
πτήσης απενεργοποιημένη.

Για να απενεργοποιήσετε τη λειτουργία πτήσης,επαναλάβετε τα παραπάνω ενώ αλλάζετε σε OFF(Ανενεργό).
όταν η οθόνη εμφανίζει τοΤο πάτημα τουσύμβολο «Next» (Επόμενο) θα μετακινήσει τησελίδα μενού στην επόμενη επιλογή οθόνης.
Η λειτουργία πτήσης προορίζεται για χρήση σεκατάσταση λειτουργίας. Η λειτουργία πτήσης θααπενεργοποιηθεί αυτόματα εάν η μονάδα αφεθεί σεκατάσταση αναμονής με τροφοδοσία από την μπαταρία.
Σημείωση: Βλ. την ενότητα «Ενεργοποίηση λειτουργίας
πτήσης» για περισσότερες πληροφορίες σχετικά με τηλειτουργία πτήσης.

ΛΕΙΤΟΥΡΓΊΑ SENSITIVITY (ΕΥΑΙΣΘΗΣΊΑ)Για να ενεργοποιήσετε τη λειτουργία, πατήστε
ενόσω το εικονίδιο CHANGE (Αλλαγή) είναι ορατό.
Η τιμή θα αλλάξει στην μπλε ένδειξη «ON» (Ενεργό).

Το FreeStyle Comfort αποστέλλεται με μιαπροεπιλεγμένη ρύθμιση λειτουργίας Sensitivity(Ευαισθησία) 2 από το εργοστάσιο και δεν αναμένεται
να απαιτεί προσαρμογή υπό κανονικές συνθήκες.
Η ρύθμιση ευαισθησίας μπορεί να προσαρμοστεί
ανάμεσα στις τιμές 1-5, όπου 1 είναι η χαμηλότερη
ευαισθησία και 5 είναι η υψηλότερη ευαισθησία.

Επιλεγμένη ενεργή λειτουργία πτήσης

Οθόνη λειτουργίας Sensitivity (Ευαισθησία) στη ρύθμιση 2Κύρια οθόνη Airplane Mode (Λειτουργία πτήσης)Για να επιβεβαιώσετε την ενεργοποίηση της λειτουργίας
ενόσω το εικονίδιο SAVEπτήσης, πατήστε
(Αποθήκευση) είναι ορατό. Η τιμή θα αλλάξει στημαύρη ένδειξη «ON» (Ενεργό).

400 - GR PN MN232-C4 L | Εγχειρίδιο χρήστη

FreeStyle ComfortΓια να ρυθμίσετε τη λειτουργία Sensitivityενόσω το εικονίδιο
(Ευαισθησία), πατήστε
CHANGE (Αλλαγή) είναι ορατό. Η τιμή θα αλλάξει
στον επόμενο μπλε αριθμό.

Λειτουργία Sensitivity (Ευαισθησία) με επιλεγμένη τη ρύθμιση 3Για να επιβεβαιώσετε τη ρύθμιση 3 στη λειτουργία
Sensitivity (Ευαισθησία), πατήστε
ενόσω τοεικονίδιο SAVE (Αποθήκευση) είναι ορατό.
Η τιμή θα αλλάξει στη μαύρη ένδειξη «3».

ΛΕΙΤΟΥΡΓΊΑ AUTOSATΤο FreeStyle Comfort αποστέλλεται με τη λειτουργία
autoSAT ενεργοποιημένη.

Κύρια οθόνη λειτουργίας autoSATΓια να απενεργοποιήσετε τη λειτουργία, πατήστε
ενόσω το εικονίδιο CHANGE (Αλλαγή) είναι ορατό.
Η τιμή θα αλλάξει στην μπλε ένδειξη «OFF» (Ανενεργό).

Για να ακυρώσετε την αλλαγή της λειτουργίας
ενόσω
Sensitivity (Ευαισθησία), πατήστε
το εικονίδιο CANCEL (Άκυρο) είναι ορατό.

Επιλεγμένη ανενεργή λειτουργία autoSATΓια να επιβεβαιώσετε την απενεργοποίηση τηςΛειτουργία Sensitivity (Ευαισθησία) με επιβεβαιωμένη τη ρύθμιση 3ενόσω το εικονίδιο
λειτουργίας autoSAT, πατήστε
SAVE (Αποθήκευση) είναι ορατό. Η τιμή θα αλλάξειστη μαύρη ένδειξη «OFF» (Ανενεργό).
Για να ακυρώσετε την ενεργοποίηση της λειτουργίας
autoSAT, πατήστε
(Άκυρο) είναι ορατό.

ενόσω το εικονίδιο CANCEL

Λειτουργία Sensitivity (Ευαισθησία) με επιβεβαιωμένητη ρύθμιση 5 (Μέγιστη)Το πάτημα τουόταν η οθόνη εμφανίζει τοσύμβολο «Next» (Επόμενο) θα μετακινήσει τη σελίδα
μενού στην επόμενη επιλογή οθόνης.

Επιβεβαιωμένη ανενεργή λειτουργία autoSATΓια να ενεργοποιήσετε τη λειτουργία autoSAT,
επαναλάβετε τα παραπάνω ενώ αλλάζετε σε ON(Ενεργό).
όταν η οθόνη εμφανίζειΤο πάτημα τουτο σύμβολο «Next» (Επόμενο) θα μετακινήσει
τη σελίδα μενού στην επόμενη επιλογή οθόνης.

PN MN232-C4 L | Εγχειρίδιο χρήστη GR - 401

FreeStyle ComfortΑντιμετώπιση προβλημάτων
ΠρόβλημαΠιθανή αιτία

ΛύσηΤο FreeStyle Comfort εμφανίζει
συναγερμό και παράγει
διαλείπον ηχητικό σήμα.

Ανατρέξτε στον πίνακα
«Καταστάσεις συναγερμού».

Ανατρέξτε στον πίνακα «Καταστάσεις συναγερμού».

Ο συμπυκνωτής οξυγόνου
δεν ενεργοποιείται.

Η μπαταρία είναι
αποφορτισμένη.

Τροφοδοτήστε τη μονάδα μέσω παροχής AC ή DC.

Δεν έχει συνδεθεί σε εξωτερική
τροφοδοσία.
Γενική δυσλειτουργία.

Διασφαλίστε ότι οι εξωτερικές συνδέσεις είναι
ασφαλείς.
Πραγματοποιήστε πλήρη κύκλο απενεργοποίησηςεπανενεργοποίησης αφαιρώντας την μπαταρία
και αποσυνδέοντας οποιαδήποτε εξωτερική
πηγή τροφοδοσίας AC ή DC. Στη συνέχεια,
απενεγκαταστήστε την μπαταρία και ο κύκλος
απενεργοποίησης-επανενεργοποίησης έχειολοκληρωθεί. Επανασυνδέστε εξωτερική πηγήτροφοδοσίας εάν είναι επιθυμητό.
Επικοινωνήστε με τον πάροχο του εξοπλισμού σαςκαι αλλάξτε σε μια άλλη πηγή οξυγόνου όπως είναι
απαραίτητο.

Όλα τα άλλα προβλήματα.

Πραγματοποιήστε πλήρη κύκλο απενεργοποίησηςεπανενεργοποίησης αφαιρώντας την μπαταρία
και αποσυνδέοντας οποιαδήποτε εξωτερική
πηγή τροφοδοσίας AC ή DC. Στη συνέχεια,
απενεγκαταστήστε την μπαταρία και ο κύκλος
απενεργοποίησης-επανενεργοποίησης έχειολοκληρωθεί. Επανασυνδέστε εξωτερική πηγήτροφοδοσίας εάν είναι επιθυμητό.
Αλλάξτε σε μια άλλη πηγή οξυγόνου που υπάρχει
διαθέσιμη και επικοινωνήστε με τον πάροχο τουεξοπλισμού σας.

Σημείωση: Ορισμένοι συναγερμοί θα προκαλέσουν απενεργοποίηση της μονάδας.
Σημείωση: Κατά τη διάρκεια πλήρους διακοπής της τροφοδοσίας, γίνεται επαναφορά του συστήματος συναγερμών
(απαλοιφή οποιωνδήποτε προηγούμενων συναγερμών) και θα εμφανιστεί συναγερμός μόνο εάν η κατάσταση
εξακολουθεί να υπάρχει όταν αποκατασταθεί η τροφοδοσία.

402 - GR PN MN232-C4 L | Εγχειρίδιο χρήστη

FreeStyle ComfortΤαξιδιωτικές πληροφορίες
Το FreeStyle Comfort πληροί τις κατευθυντήριες
οδηγίες της FAA για χρήση επί εμπορικού
αεροσκάφους. Πριν από οποιοδήποτε ταξίδι
(με αεροπλάνο, τρένο, πλοίο κ.λπ.) πρέπει να:• Διασφαλίσετε ότι ο συμπυκνωτής είναι σε καλήκατάσταση λειτουργίας.
• Επικοινωνήστε με την εταιρεία με την οποία
θα ταξιδέψετε τουλάχιστον 48 ώρες πριν από τηναναχώρηση για να ενημερώσετε ότι θα ταξιδέψετε
με ένα FreeStyle Comfort.
Συνιστάται να έχετε πάντα μαζί σας επιπλέον
μπαταρίες στα ταξίδια. Για αεροπορικά ταξίδια,προβλέψτε αρκετή διάρκεια ζωής της μπαταρίας
για τον χρόνο στο έδαφος (check in πριν την πτήση,έλεγχο ασφάλειας, κ.λπ.) και ενδιάμεσες στάσεις.
Οι περισσότερες αεροπορικές εταιρείες απαιτούν ναέχετε επαρκή αριθμό πλήρως φορτισμένων πακέτων
μπαταρίας για την τροφοδοσία της συσκευής γιατουλάχιστον 150% της μέγιστης αναμενόμενηςδιάρκειας του ταξιδιού (χρόνοι πτήσης, χρόνος στοέδαφος πριν και μετά την πτήση, ανταποκρίσεις καιαπρόβλεπτες καθυστερήσεις).
Σημείωση: Εάν ταξιδεύετε αεροπορικώς, συνιστάται
να μεταφέρετε τις επιπλέον μπαταρίες σε μια ασφαλή
τσάντα μεταφοράς.

ΚΑΝΟΝΙΣΜΟΣ FAA

1. Ενεργοποιήστε τον POC χρησιμοποιώντας
τροφοδοσία από μπαταρία ΜΟΝΟ, όταν δεν είναι
συνδεδεμένη εξωτερική πηγή τροφοδοσίας.

2. Ενεργοποιήστε τη λειτουργία πτήσης
χρησιμοποιώντας μία από τις ακόλουθες μεθόδους:α. Ακολουθήστε τις οδηγίες για την ενεργοποίησητης λειτουργίας πτήσης στην ενότητα «Οδηγίες
πλοήγησης μενού».

β. Πατήστε και κρατήστε πατημένα και τα δύοκουμπιά Πάνω και Κάτω για 5 δευτερόλεπταμέχρι να ακούσετε ένα μακρύ ηχητικό σήμα καινα εμφανιστεί το σύμβολο αεροσκάφους.

Οι εφεδρικές μπαταρίες πρέπει να προστατεύονταιατομικά ώστε να αποφεύγονται τα βραχυκυκλώματα,τοποθετώντας τις στην αρχική συσκευασία πώλησης
ή μονώνοντας τους ακροδέκτες με άλλον τρόπο, π.χ.
καλύπτοντας με ταινία τους εκτεθειμένους ακροδέκτεςή τοποθετώντας κάθε μπαταρία σε ένα ξεχωριστό
πλαστικό σακουλάκι ή προστατευτική θήκη, και ναμεταφέρονται μόνο στις χειραποσκευές.

ΕΝΕΡΓΟΠΟΙΗΣΗ ΛΕΙΤΟΥΡΓΙΑΣ ΠΤΗΣΗΣΤροφοδοσία από μπαταρία
Μπορεί να χρησιμοποιηθεί POC (φορητός
συμπυκνωτής οξυγόνου) σε αεροσκάφος σύμφωνα μετις κανονικές διαδικασίες λειτουργίας χρησιμοποιώντας
τροφοδοσία από μπαταρία ΜΟΝΟ, όταν δεν είναι
συνδεδεμένη εξωτερική πηγή τροφοδοσίας
Εξωτερική τροφοδοσίαΕπί του αεροσκάφους, όταν είναι διαθέσιμη πρίζα
ηλεκτρικού ρεύματος, και η θέση της είναι κατάλληληγια χρήση ως πηγή τροφοδοσίας του POC, μπορεί ναχρησιμεύσει ως εφεδρική στις μπαταρίες του POC.
Στην περίπτωση αυτή, πρέπει να τηρηθεί η ακόλουθησυνιστώμενη διαδικασία όσον αφορά τη μετάβαση απότην μπαταρία στην παροχή ηλεκτρικού ρεύματος τουαεροσκάφους:3. Όταν εμφανιστεί το σύμβολο αεροσκάφους,συνδέστε τον POC.

Σημείωση: Λόγω περιορισμών των αεροσκαφών,οι μπαταρίες δεν φορτίζονται σε λειτουργία πτήσης.

PN MN232-C4 L | Εγχειρίδιο χρήστη GR - 403

FreeStyle Comfort
Παρελκόμενα

ΤΟΠΟΘΕΤΗΣΗ ΤΗΣ ΜΟΝΑΔΑΣ ΣΤΗΝΤΣΑΝΤΑ ΜΕΤΑΦΟΡΑΣ

Για σωστή απόδοση και ασφάλεια, χρησιμοποιείτε μόνοαυτά τα παρατιθέμενα παρελκόμενα που παρέχονταιαπό την CAIRE μέσω του παρόχου οξυγόνου σας. Ηχρήση παρελκομένων που δεν παρατίθενται παρακάτω
θα μπορούσε να επηρεάσει αρνητικά την απόδοση ή/καιασφάλεια του συμπυκνωτή.

1. Τοποθετήστε τη μονάδα στο κάτω μέρος τηςτσάντας, ευθυγραμμίζοντας σωστά τα φίλτρα και ταανοίγματα μπαταρίας.

Τυπικά παρελκόμενα του FreeStyle ComfortΤσάντα μεταφοράς

MI396-1Σακίδιο πλάτης

MI459-1Ιμάντας ώμουMI397-1Τροφοδοτικό AC (συμπεριλαμβάνονταιόλα τα καλώδια)PW036-1SΚαλώδιο ρεύματος DC

CD041-1Εξωτερικό φίλτρο χονδρών σωματιδίων (δεξιά)FI226-1Εξωτερικό φίλτρο χονδρών σωματιδίων (αριστερά)FI227-1

2. Κλείστε καλά το φερμουάρ γύρω από το κάτω μέροςτης θήκης.

Διαθέσιμα προαιρετικά εξαρτήματα
Πρόσθετο πακέτο μπαταρίας
8 στοιχείων

BT034-1SΣυστοιχία μπαταριών 16 στοιχείων

BT035-1SΕπιτραπέζιος φορτιστής πακέτου
μπαταρίας

BT036-1Τσάντα παρελκομένων «carry-all»

MI372-2Σημείωση: Μπορεί να είναι διαθέσιμες πρόσθετες
επιλογές για καλώδια τροφοδοσίας ειδικά για τηχώρα στις περιπτώσεις που αναφέρονται παραπάνω.
Επικοινωνήστε με την CAIRE ή με τον πάροχο οξυγόνου
σας εάν χρειάζεται να παραγγείλετε εναλλακτικές
επιλογές.

Οδηγίες τσάντας μεταφοράς
Για να τοποθετήσετε και να ασφαλίσετε το FreeStyle
Comfort στην τσάντα μεταφοράς, ακολουθήστε τιςπαρακάτω οδηγίες.

404 - GR PN MN232-C4 L | Εγχειρίδιο χρήστη

ΤΟΠΟΘΕΤΗΣΗ ΤΟΥ ΙΜΑΝΤΑ ΩΜΟΥ
1. Ο ιμάντας ώμου μπορεί να προσαρτηθεί είτεαπευθείας στη μονάδα είτε στην τσάντα μεταφοράς.
2. Συνδέστε τον ιμάντα με το κλιπ στον δακτύλιο πουβρίσκεται στη μονάδα ή στην τσάντα.

FreeStyle ComfortΔιάθεση και επαναχρησιμοποίηση
Πάντοτε να επιστρέφετε το FreeStyle Comfort,
συμπεριλαμβανομένων όλων των εξαρτημάτων, στονπρομηθευτή εξοπλισμού σας για σωστή απόρριψη.
Μπορείτε επίσης να επικοινωνήσετε με τα τοπικά
γραφεία της πόλης σας για οδηγίες σχετικά με τησωστή απόρριψη της μπαταρίας. Η μονάδα μπορεί
να επαναχρησιμοποιηθεί από άτομο σε άτομο, αλλάπρέπει να επιστραφεί στον προμηθευτή εξοπλισμού
για επανεπεξεργασία.

ΑΗΗΕ και RoHSΑυτό το σύμβολο υπενθυμίζει στους κατόχους
του εξοπλισμού να τον επιστρέψουν σε εγκαταστάσεις
ανακύκλωσης στο τέλος της ωφέλιμης ζωής του,σύμφωνα με την Οδηγία για τα Απόβλητα Ηλεκτρικού
και Ηλεκτρονικού Εξοπλισμού (ΑΗΗΕ).

Υλικά σε άμεση επαφή με τονχειριστή• Περίβλημα συμπυκνωτή: PC/ABS• Πίνακας ελέγχου συμπυκνωτή: Πολυεστέρας EBG7ή ισοδύναμο• Έξοδος οξυγόνου, ρύγχος: Αλουμίνιο• Ετικέτα μονάδας: Lexan• Σύνδεσμοι καλωδίου: Πολυανθρακικό/χλωριούχο
βινύλιο• Καλώδιο ρεύματος: PVC, μέταλλο• Τροφοδοτικό: Lexan 940 (πολυανθρακικό)• Συστοιχία μπαταρίας: PC/ABS• Ετικέτες συστοιχίας μπαταρίας, τροφοδοτικού:Πολυεστερική μεμβράνη• Θήκη μεταφοράς συμπυκνωτή: Cordura, νεοπρένιο,επένδυση από νάιλον• Ιμάντας: Πολυπροπυλένιο, ακετύλιο, βινύλιο,Cordura, νεοπρένιο

Τα προϊόντα μας συμμορφώνονται με την οδηγία γιατον περιορισμό των επικίνδυνων ουσιών (RoHS).
Δεν περιέχουν περισσότερο από ιχνοποσότητες
μολύβδου ή άλλων επικίνδυνων υλικών.
ΠΡΟΣΟΧΗ: Για σωστή απόρριψη, επικοινωνήστε με τον πάροχο του εξοπλισμού
σας ή με την αρμόδια υπηρεσία τοπικής
αυτοδιοίκησης για οδηγίες απόρριψης
ΠΡΟΣΟΧΗ: Να απορρίπτετε πάντα το μοριακό
κόσκινο σύμφωνα με όλους τους ομοσπονδιακούς,πολιτειακούς και τοπικούς κανονισμούς
περιβαλλοντικού ελέγχου.

PN MN232-C4 L | Εγχειρίδιο χρήστη GR - 405

FreeStyle ComfortΔοκιμές ηλεκτρομαγνητικές συμβατότητας (EMC)Οδηγία και δήλωση του κατασκευαστή – ηλεκτρομαγνητικές εκπομπές
Το FreeStyle Comfort προορίζεται για χρήση στο ηλεκτρομαγνητικό περιβάλλον που προσδιορίζεται παρακάτω.
Ο αγοραστής ή ο χρήστης του FreeStyle Comfort θα πρέπει να διασφαλίσει ότι το σύστημα χρησιμοποιείται
σε ανάλογο περιβάλλον.
Δοκιμή εκπομπών
Συμμόρφωση
Ηλεκτρομαγνητικό περιβάλλον – οδηγίες
Το FreeStyle Comfort χρησιμοποιεί ενέργεια ραδιοσυχνοτήτων
Εκπομπές(RF) μόνο για την εσωτερική λειτουργία του. Συνεπώς, οι εκπομπές
ραδιοσυχνοτήτων (RF)Ομάδα 1ραδιοσυχνοτήτων της συσκευής είναι πολύ χαμηλές και δεν είναι
πιθανό να προκαλέσουν παρεμβολή σε κοντινό ηλεκτρονικόCISPR 11εξοπλισμό.
Εκπομπές
ραδιοσυχνοτήτων (RF)Κατηγορία B
Το FreeStyle Comfort είναι κατάλληλο για χρήση σε όλους τουςCISPR 11χώρους, συμπεριλαμβανομένων των οικιακών εγκαταστάσεων
Εκπομπές αρμονικών
και εκείνων που είναι απευθείας συνδεδεμένοι στο δημόσιο
Κατηγορία A
IEC 61000-3-2δίκτυο παροχής χαμηλής τάσης, που τροφοδοτεί κτίρια πουΔιακυμάνσεις τάσης/χρησιμοποιούνται για οικιακούς σκοπούς.
Συμμορφώνεταιεκπομπές αναλαμπής
IEC 61000-3-3

406 - GR PN MN232-C4 L | Εγχειρίδιο χρήστη

FreeStyle ComfortΟδηγίες και δήλωση κατασκευαστή – ηλεκτρομαγνητική ατρωσία

Το FreeStyle Comfort προορίζεται για χρήση στο ηλεκτρομαγνητικό περιβάλλον που προσδιορίζεται παρακάτω.
Ο αγοραστής ή ο χρήστης του FreeStyle Comfort θα πρέπει να διασφαλίσει ότι το σύστημα χρησιμοποιείται
σε ανάλογο περιβάλλον.
Δοκιμή ατρωσίας
Επίπεδο δοκιμής
Επίπεδο συμμόρφωσης Ηλεκτρομαγνητικό περιβάλλον – οδηγίες
IEC 60601
Τα δάπεδα θα πρέπει να είναι κατασκευΗλεκτρομαγνητικό
±8 kV επαφή
±8 kV επαφή
ασμένα από ξύλο, τσιμέντο ή κεραμικά
περιβάλλον – οδηγίες ±15 kV εκφόρτιση στονπλακίδια. Εάν τα δάπεδα είναι επιστρωμέ±15 kV εκφόρτιση
να με συνθετικό υλικό, η σχετική υγρασία
στον αέρα
IEC 61000-4-2αέραπρέπει να είναι τουλάχιστον 30%.
Η ποιότητα των κυρίων γραμμών
±2 kV για γραμμές
±2 kV για γραμμές
τροφοδοσίας πρέπει να είναι ισοδύναμη
τροφοδοσίας ρεύματοςΤαχεία ηλεκτρική
τροφοδοσίας ρεύματος
με εκείνη μιας τυπικής επαγγελματικής
πάροδος/ριπήεγκατάστασης υγειονομικής περίθαλψης
IEC 61000-4-4±1 kV για γραμμές
και ενός τυπικού περιβάλλοντος
Δεν ισχύει
εισόδου/εξόδου
περίθαλψης κατ' οίκον.
±1 kV γραμμή(ές) προςγραμμή(ές)ΥπέρτασηIEC 61000-4-5±2 kV τρόπος λειτουργίας
κοινού σήματος στις
±1 kV γραμμή(ές)γραμμές AC
προς γραμμή(ές)±1 kV διαφορικός
τρόπος λειτουργίας
στις γραμμές ACΗ ποιότητα των κυρίων γραμμών τροφοδοσίας πρέπει να είναι ισοδύναμη με εκείνη
μιας τυπικής επαγγελματικής εγκατάστασης υγειονομικής περίθαλψης και ενόςτυπικού περιβάλλοντος περίθαλψης κατ'οίκον.

±2 kV τρόπος λειτουργίας
κοινού σήματος στις
γραμμές I/O (εισόδου/εξόδου)0% UT για 0,5 κύκλο
0% UT για 0,5 κύκλο
(0°, 45°, 90°, 135°, 180°, (0°, 45°, 90°, 135°, 180°, Η ποιότητα των κυρίων γραμμών
τροφοδοσίας πρέπει να είναι ισοδύναμη
225°, 270°, 315°)
225°, 270°, 315°)με εκείνη μιας τυπικής επαγγελματικής
Πτώσεις τάσης,εγκατάστασης υγειονομικής περίθαλψης
σύντομες διακοπές
0% UT για 1 κύκλο (0°)
0% UT για 1 κύκλο (0°)και ενός τυπικού περιβάλλοντος
και διακυμάνσεις
περίθαλψης κατ' οίκον. Εάν ο χρήστης
τάσης στις γραμμές
του FreeStyle Comfort απαιτεί συνεχή
70% UT (30% βύθιση
70% UT (30% βύθιση
παροχής τροφολειτουργία κατά τη διάρκεια διακοπών
)
για
25/30
)
για
25/30
στην
U
στην
U
δοσίας εισόδου
T
Tρεύματος, συνιστάται η τροφοδοσία
κύκλους (0°)κύκλους (0°)
IEC 61000-4-11
του FreeStyle Comfort να γίνεται απότροφοδοτικό αδιάλειπτης λειτουργίας
0% UT για 250/300
0% UT για 250/300
ή μπαταρία.
κύκλους (0°)κύκλους (0°)Τα μαγνητικά πεδία συχνότητας ρεύματοςΜαγνητικό πεθα πρέπει να βρίσκονται σε επίπεδαδίο συχνότητας
χαρακτηριστικά μιας τυπικής τοποθεσίας
3A/m
3A/m
ισχύος (50/60 Hz)σε μια επαγγελματική εγκατάσταση
IEC 61000-4-8υγειονομικής περίθαλψης και ένα τυπικό
περιβάλλον περίθαλψης κατ' οίκον.
ΣΗΜΕΙΩΣΗ UT είναι η τάση κύριας παροχής τάσης δικτύου (AC) πριν την εφαρμογή του επιπέδου ελέγχου.

ΣΗΜΕΙΩΣΗ Σε ορισμένες περιπτώσεις, το FreeStyle Comfort μπορεί να κάνει επαναφορά μετά από σημαντική
ηλεκτροστατική εκφόρτιση και μπορεί να απαιτεί πλήρη κύκλο απενεργοποίησης-επανενεργοποίησης για να συνεχίσει
την κανονική λειτουργία. Βλ. την ενότητα «Αντιμετώπιση προβλημάτων» για περισσότερες λεπτομέρειες.

PN MN232-C4 L | Εγχειρίδιο χρήστη GR - 407

FreeStyle ComfortΟδηγίες και δήλωση κατασκευαστή – ηλεκτρομαγνητική ατρωσία

Το FreeStyle Comfort προορίζεται για χρήση στο ηλεκτρομαγνητικό περιβάλλον που προσδιορίζεται παρακάτω.
Ο αγοραστής ή ο χρήστης του FreeStyle Comfort θα πρέπει να διασφαλίσει ότι το σύστημα χρησιμοποιείται
σε ανάλογο περιβάλλον.
Επίπεδο συμμόρφωσης Ηλεκτρομαγνητικό περιβάλλον – οδηγίες
Δοκιμή ατρωσίας Επίπεδο δοκιμής
IEC 60601
Αγόμενες
10 Vrms
10 VrmsΟ φορητός και κινητός εξοπλισμός
ραδιοσυχνότητες 150 kHz έως 80 MHzεπικοινωνιών με ραδιοσυχνότητες δενπρέπει να χρησιμοποιείται πλησιέστερα
IEC 61000-4-6σε οποιοδήποτε εξάρτημα του FreeStyle
Comfort, συμπεριλαμβανομένων τωνκαλωδίων, από τη συνιστώμενη απόσταση10 V/m
10 V/mδιαχωρισμού που υπολογίζεται από τηνΑκτινοβολούμενες 80 MHz έως 2700 MHzεξίσωση που εφαρμόζεται στη συχνότητα
ραδιοσυχνότητες
του πομπού.
IEC 61000-4-3Συνιστώμενη απόσταση διαχωρισμού
= 1.2 Pd
= 1.2 P = 80 MHz έως 800 MHzd= 1.2 P = 800 MHz έως 2700 MHzdόπου P είναι η μέγιστη τιμή ισχύος εξόδου
του πομπού σε watt (W) σύμφωνα με τονκατασκευαστή του πομπού και d είναιη συνιστώμενη ενδιάμεση απόσταση σεμέτρα (m).
Η ισχύς πεδίου από σταθερούς πομπούς
ραδιοσυχνοτήτων, όπως καθορίζεται απόμία ηλεκτρομαγνητική μελέτη του χώρου, απρέπει να είναι χαμηλότερη από το επίπεδο
συμμόρφωσης σε κάθε εύρος συχνοτήτων. βΕνδέχεται να εμφανιστεί παρεμβολή στοπεριβάλλον εξοπλισμού που επισημαίνεταιμε το ακόλουθο σύμβολο:ΣΗΜΕΙΩΣΗ 1 Στα 80 MHz και τα 800 MHz, ισχύει το υψηλότερο εύρος συχνοτήτων.
ΣΗΜΕΙΩΣΗ 2 Αυτές οι οδηγίες μπορεί να μην ισχύουν σε όλες τις καταστάσεις. Η διάδοση των ηλεκτρομαγνητικών
κυμάτων επηρεάζεται από την απορρόφηση και την αντανάκλαση από δομές, αντικείμενα και ανθρώπους.
αΗ ισχύς πεδίου από σταθερούς πομπούς, όπως σταθμοί βάσης για ραδιοτηλέφωνα (κινητά/ασύρματα) και επίγεια κινητά
ραδιόφωνα, ερασιτεχνικές ραδιοφωνικές εκπομπές, ραδιοφωνικές αναμεταδόσεις AM και FM, καθώς και τηλεοπτικές
αναμεταδόσεις, δεν μπορεί να προβλεφθεί θεωρητικά με ακρίβεια. Για να αξιολογήσετε το ηλεκτρομαγνητικό περιβάλλον
λόγω των σταθερών πομπών ραδιοσυχνοτήτων, θα πρέπει να εξετάζεται το ενδεχόμενο διερεύνησης της θέσης.
Εάν η μετρημένη ισχύς πεδίου στην τοποθεσία όπου χρησιμοποιείται το FreeStyle Comfort υπερβαίνει το παραπάνω
εφαρμοζόμενο επίπεδο συμμόρφωσης ραδιοσυχνοτήτων, το FreeStyle Comfort πρέπει να τεθεί υπό παρακολούθηση μεσκοπό την επαλήθευση της κανονικής λειτουργίας. Εάν παρατηρηθεί μη κανονική απόδοση, ενδέχεται να απαιτούνται
πρόσθετα μέτρα, όπως η αλλαγή προσανατολισμού ή θέσης του FreeStyle Comfort.
βΣτο εύρος συχνοτήτων 150 kHz έως 80 MHz, η ισχύς πεδίου πρέπει να είναι μικρότερη από 10 V/m.

408 - GR PN MN232-C4 L | Εγχειρίδιο χρήστη

FreeStyle ComfortΣυνιστώμενες αποστάσεις διαχωρισμού μεταξύ φορητού και κινητού εξοπλισμού
επικοινωνιών με ραδιοσυχνότητες και του FreeStyle Comfort

Το FreeStyle Comfort προορίζεται για χρήση σε ηλεκτρομαγνητικό περιβάλλον στο οποίο οι διαταραχές από εκπομπές
ραδιοσυχνοτήτων είναι ελεγχόμενες. Ο αγοραστής ή ο χρήστης του FreeStyle Comfort μπορεί να βοηθήσει στην πρόληψη
των ηλεκτρομαγνητικών παρεμβολών, διατηρώντας μια ελάχιστη απόσταση μεταξύ του φορητού και κινητού εξοπλισμού
επικοινωνιών με ραδιοσυχνότητες (πομπών) και του FreeStyle Comfort, όπως συνιστάται παρακάτω, σύμφωνα με τημέγιστη ισχύ εξόδου του εξοπλισμού επικοινωνιών.
Ονομαστική τιμήΑπόσταση διαχωρισμού ανάλογα με τη συχνότητα του πομπού
μέγιστης ισχύος εξόδουmτου πομπού
80 MHz έως 800 MHz
800 MHz έως 2,5 GHz
150 kHz έως 80 MHz= 1.2 P= 1.2 Pd = 1.2 P
d
d
W
0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
Για πομπούς με ονομαστική μέγιστη ισχύ εξόδου που δεν παρατίθεται παραπάνω, η συνιστώμενη απόστασηδιαχωρισμού d σε μέτρα (m) μπορεί να προσδιοριστεί με χρήση της εξίσωσης που εφαρμόζεται στη συχνότητα τουπομπού, όπου P είναι η μέγιστη τιμή ισχύος εξόδου του πομπού σε watt (W), σύμφωνα με τον κατασκευαστή του πομπού.
ΣΗΜΕΙΩΣΗ 1 Στα 80 MHz και 800 MHz, ισχύει η απόσταση διαχωρισμού για το υψηλότερο εύρος συχνοτήτων.
ΣΗΜΕΙΩΣΗ 2 Αυτές οι οδηγίες μπορεί να μην ισχύουν σε όλες τις καταστάσεις. Η διάδοση των ηλεκτρομαγνητικών
κυμάτων επηρεάζεται από την απορρόφηση και την αντανάκλαση από δομές, αντικείμενα και ανθρώπους.

Ταξινόμηση

Τύπος προστασίας από ηλεκτροπληξία:Κατηγορία II	Η προστασία από ηλεκτροπληξία
			επιτυγχάνεται με διπλή μόνωση.
Βαθμός προστασίας από ηλεκτροπληξία:Εξοπλισμός 	Τύπου BF που παρέχει συγκεκριμένο
		βαθμό προστασίας από ηλεκτροπληξία σχετικά με		1) το επιτρεπόμενο ρεύμα διαρροής,		2) την αξιοπιστία της σύνδεσης προστατευτικής
γείωσης (εάν υπάρχει).
		Δεν προορίζεται για άμεση καρδιακή εφαρμογή.
Ανεξάρτητες δοκιμές για το πρότυπο για ιατρικές
ηλεκτρικές συσκευές:	Δοκιμάστηκε για συμμόρφωση με• IEC 60601-1 έκδοση 3.1: Ιατρικές ηλεκτρικές
συσκευές – Μέρος 1 Γενικές απαιτήσεις για τηνασφάλεια.
• CAN/CSA C22.2 Αρ. 60601-1-8 έκδοση 2.1,Ιατρικές ηλεκτρικές συσκευές – Μέρος 1:Γενικές απαιτήσεις για την ασφάλεια
Προστασία έναντι πιθανών ηλεκτρομαγνητικών ή άλλων
παρεμβολή μεταξύ του εξοπλισμού και άλλων συσκευών.
	• Δοκιμάστηκε για συμμόρφωση με EN 60601-1-2/
IEC 60601-1-2: 4η έκδοση.

	• Δοκιμάστηκε για συμμόρφωση με RTCA/DO160
	Ενότητα 21 Κατηγορία M.
• CISPR 11 / EN 55011 Ομάδα 1 Κατηγορίας Β,	«Βιομηχανικές, επιστημονικές καιιατροτεχνολογικές (ISM) συσκευές»• FCC Μέρος 15, Τμήμα B – Κατηγορία B	Ακούσιοι ακτινοβολητές
IP22 – Προστασία κατά της εισόδου δακτύλων
ή παρόμοιων αντικειμένων με διάμετρο μεγαλύτερη
από 12,5 mm. Η κάθετη πτώση σταγόνων νερού δενθα έχει επιβλαβείς επιπτώσεις όταν το περίβλημαβρίσκεται υπό γωνία κλίσης 15° από την κανονική
του θέση.
Επιτρεπόμενη μέθοδος καθαρισμού και ελέγχου τωνλοιμώξεων: Παρακαλώ ανατρέξτε στην ενότητα
«Καθαρισμός, φροντίδα και συντήρηση ρουτίνας»του παρόντος εγχειριδίου χρήστη του FreeStyle.
Βαθμός ασφάλειας της εφαρμογής υπό την παρουσία
εύφλεκτων αναισθητικών αερίων:	Εξοπλισμός που δεν είναι κατάλληλος για τέτοια
εφαρμογή.
Τρόπος λειτουργίας: Συνεχής λειτουργία.

PN MN232-C4 L | Εγχειρίδιο χρήστη GR - 409

FreeStyle Comfort
Kullanıcı Kontrolleri ve Sistem Durumu Göstergeleri
ISO 7000
Operatör el kitabı; çalıştırma
talimatları. Yön. No. 1641
Ülke ve üretim tarihi. "CC", üretildiği
ülkenin iki harfli ülke kodunu
tanımlar. Üretim tarihi YYYY-AA-GG
formatındadır. Yön. No. 6049

ISO 7010: Grafik sembolleri — Güvenlik renkleri
ve güvenlik işaretleri — Kayıtlı güvenlik işaretleri
Kullanıcı el kitabı okunmalıdır.
Yön. No. M002
Açık alev, yangın ve kıvılcımdan
uzak tutun. Yakınında açık alev
kaynağı bulundurmak ve sigara içmek
yasaktır. Yön. No. P003
Ünitenin yakınındayken veya üniteyiçalıştırırken sigara içmeyin. Yön.
No. P002Yağmurdan koruyun, kuru tutun.
Yön. No. 0626
Sayıya göre yığınlama sınırı; burada"n", izin verilen maksimum birimsayısını gösterir. Yön. No. 2403
Katalog Numarası. Yön. No. 2493

Uyarı. Yön. No. W001
IEC 60417
BF Tipi uygulamalı parça (elektrikçarpmasına karşı koruma derecesi).
Yön. No. 5333

Seri Numarası. Yön. No. 2498
Saklama veya çalışma sıcaklığı
aralığı. Yön. No. 0632
Saklama nemi aralığı. Yön. No. 2620
Atmosfer basıncı sınırlaması.
Yön. No. 2621
Üreticinin adı ve adresi. Yön.
No. 3082
Üst kısmıdır. Yön. No. 0623

Sınıf II Ekipman, İki Kat Yalıtımlı
Yön. No. 5172
AB Yönetmeliği 2017/745
Avrupa Topluluğundaki yetkili
temsilciÜrün benzersiz cihaz tanımlayıcısı
(UDI) etiketinin üzerinde CE####
sembolü varsa cihaz tıbbi cihazlarailişkin gereksinimlerle uyumludur.
CE#### sembolü, onaylanmış kuruluş
numarasını gösterir.
Ek Semboller
Yanıcı materyallerden, yağdan
ve gresten uzak tutun.

Dikkat, kırılır. Yön. No. 0621

Parçalarına ayırmayın.

Tehlikeli maddeler içerir. Yön.
No. 3723

Akışı artırın.

İthalatçı. Yön. No. 3725
Akışı azaltın.
O2 Çıkışı

Menü düğmesi.
Güç anahtarı. Yön. No. 5009
RTCA/
DO-160Bölüm 21
M Kategorisi
ile Uyumlu

410 - TR PN MN232-C4 L | Kullanıcı El Kitab

Bu POC'nin üreticisi, bu cihazın
geçerli tüm FAA kabul kriterlerine
ve uçakta kullanıma uygun olduğunu
belirlemiştir.

FreeStyle Comfort
Bu POC'nin üreticisi, bu cihazın
geçerli tüm FAA kabul kriterlerine
ve uçakta kullanıma uygun olduğunu
belirlemiştir.
Geri Dönüşüm İşareti

IEC 60601-1
Sızdırmaz
Konsey Direktifi 2012/19/AB
WEEE
Bu sembol, ekipmanın sahiplerine
ekipmanı kullanım ömrü sonunda Atık
Elektrikli ve Elektronik Ekipmanlar(WEEE) Direktifine uygun olarak bir
geri dönüşüm tesisine iade etmelerini
hatırlatmaya yöneliktir. Ürünlerimiz,
Tehlikeli Maddelerin Kısıtlanması(RoHS) direktifine uyar. Eser
miktardan fazla kurşun veya başka
tehlikeli malzeme içermezler.

İsviçre'deki yetkili temsilci (ADD)
Cihaz, onaylanmış kuruluş numarasını
belirten UKCA#### ile gösterildiği
gibi UKCA işaretini taşıyorsa bu
cihaz UKCA yönetmeliklerine
uygundur.
21 CFR 801.15
Federal yasalar uyarınca bu cihazyalnızca bir hekim tarafından veya
hekimin talimatıyla satılabilir.

IEC 60601-1-8Düşük Öncelikli Teknik AlarmETL Sertifikası
Hem ABD hem de Kanada
piyasaları için ilgili ABD ve
Kanada standartlarına uygun olarak
onaylanmıştır
FCC kimliğini içerir: WAP2001
FCC Bildirimi
Bu cihaz CYBLE-022001-00 (Johnson Technology
firmasının 2450AT18B100 no’lu anteni dahil)
içerebilir ve FCC Kuralları'nın 15. bölümü ile
uyumludur. Cihaz, DA00-1407 sayılı FCC kamu
bilgilendirmesinde ayrıntılarıyla açıklanan şekilde
onaylanmış modüler verici gereksinimlerinikarşılamaktadır. Vericinin çalışması şu ikikoşula tabidir: (1) Cihaz, zararlı parazite neden
olmamalıdır ve (2) Bu cihaz, çalışmasını
istenmeyen bir şekilde etkileyebilecek parazitler
dahil aldığı tüm parazitleri kabul etmelidir.
Ürün, ABD’de ve tüm dünyada geçerli bir
veya daha fazla patent kapsamında olabilir.
Geçerli patentlerin listesini görmek için lütfen
aşağıdaki web sitemizi ziyaret edin. Pat:www.caireinc.com/corporate/patents

ISO 15223-1
Tıbbi Cihaz. Yön. No. 5.7.7
Benzersiz cihaz tanımlayıcı no. 5.7.10
A S/NZS 4417.1, AS/NZS 4417.2
ACMA Düzenleyici Uygunlukİşareti – ACMA yönetmeliklerine(ADD) uygunluğu gösterir
Kablosuz TeknolojiBağlantı türleri

SSP, iAP2, GATT

Sıklık2402 - 2480 MHz

Maks. RF gücü çıkışı

+4 dBmÇalışma aralığı

10 m (Sınıf 2)

Makinenin, çalıştığı sırada gövdeden en az 1,1 cm(0,43 inç) uzakta olması önerilir. Maskeler, nefes
boruları veya aksesuarlar için geçerli değildir.

Ekipman Üzerinde Kullanılan Grafik Sembolleri
IEC 60417
Doğru Akım Yön. No. 5031
Alternatif Akım Yön. No. 5032

PN MN232-C4 L | Kullanıcı El Kitab TR - 411

FreeStyle Comfort
FreeStyle Comfort Ürününüzü Kutusundan Çıkarma
Listelenen tüm bileşenlerin
ambalajda mevcut olduğunu
doğrulayın. Eksik bir parça
varsa oksijen sağlayıcınızla
hemen iletişime geçin.

Standart Aksesuarlar• Taşınabilir Oksijen Yoğunlaştırıcı• Taşıma Çantası• Omuz Askısı• 8 Hücreli Pil Takımı• AC Güç Kaynağı
• AC Güç Kablosu• DC Güç Kaynağı
• Kullanıcı El Kitabı

Opsiyonel Aksesuarlar• Ek 8 Hücreli Pil Takımı• 16 Hücreli Pil Takımı• Masaüstü Şarj Cihazı• Tüm Parçalar için AksesuarÇantasıFreeStyle Comfort Ürününüzü Tanıma
FreeStyle Comfort ve kullanıcı arayüzünün temel özelliklerini tanıyın.

Önden Görünüm
Askı Ek Parçası

Kullanıcı
ArayüzüOksijenÇıkışı
Hava Çıkışı

Hava Girişi/Filtreler
Hava Girişi/Filtreler
Güç Portu
Priz Tırnağı
Pil Takımı
Pil Takımını
Çıkarma Düğmesi
412 - TR PN MN232-C4 L | Kullanıcı El Kitab

FreeStyle ComfortÜstten Görünüm
Görüntü EkranıAkışı Artırma

Güç Düğmesi

Menü Düğmesi

Akışı Azaltma

Pilde Görüntü EkranıAkış Ayarı

Pil Ekranı

autoSAT Göstergesi
Alarm Ekranı

Kalan Pil
Yüzdesi

Bolus Verilme
Göstergesi

AC/DC Güçte Görüntü EkranıAkış Ayarı

Alarm Ekranı
Alanı

Pil EkranıŞarj Göstergesi

PN MN232-C4 L | Kullanıcı El Kitab TR - 413

FreeStyle Comfort
Oksijen Yoğunlaştırıcı Nedir?
Soluduğumuz hava yaklaşık %21 oksijen, %78 azot
ve %1 oranında diğer gazlardan oluşur. FreeStyle®
Comfort®, odadaki havayı hava girişleri yoluyla
içine çeker. Hava daha sonra moleküler elek adı
verilen tutucu bir maddeden geçirilir. Bu madde,oksijeni azottan ayırır ve yalnızca oksijeningeçmesine izin verir. Bunun sonucunda, kullanıcıya
yüksek yoğunlukta oksijen akışı sağlanır.
FreeStyle Comfort, gelişmiş oksijen yoğunlaştırıcı
ve oksijen koruma teknolojilerini birleştirerek
maksimum taşınabilirlik ve hareketliliğe olanak
tanıyan taşınabilir bir cihazdır. Gelişmiş puls akışı
sunumu, kullanıcının nefes alışını hızla algılar ve
alınan her nefesin başlangıcında bir puls oksijensağlar. Puls akışı sunumu, evin dışında yapılan
faaliyetler ve geçirilen zaman için idealdir. Sürekliakışın aksine, oksijeni muhafaza eder ve pil ömrünü
uzatır. FreeStyle Comfort, omuz askısı ile taşınacakkadar hafiftir. AC (duvar prizi), DC (taşıt elektriği)veya pil gücüyle çalıştırılabilir.

Kullanım Amacı
CAIRE FreeStyle Comfort OksijenYoğunlaştırıcının ek oksijen uygulaması içinkullanılması amaçlanmıştır. Bu cihaz yaşam
desteğine yönelik olmadığı gibi hasta izlemeözelliklerine de sahip değildir.

Hekiminiz Neden Oksijen Reçete Etti?
Birçok kişi çeşitli kalp, akciğer ve diğer solunum
hastalıklarından şikayetçidir. Bu tür hastalıklardanşikayetçi olan çok sayıda kişi evde, seyahatsırasında veya evin dışında gündelik faaliyetlere
katılırken ek oksijen terapisinin faydasını görebilir.
Oksijen, soluduğumuz ortam havasının %21'inioluşturan bir gazdır ve vücudumuzun düzgün
şekilde çalışması için sabit bir oksijen kaynağına
ihtiyacı vardır. Hekiminiz, solunum durumunuza
fayda sağlaması için bir akışı veya ayarı reçeteetmiştir.

414 - TR PN MN232-C4 L | Kullanıcı El Kitab

Oksijen bağımlılık yapmayan bir ilaç olsa daonaysız oksijen tedavisi tehlikeli olabilir. Bu
oksijen yoğunlaştırıcıyı kullanmadan önce hekimedanışmanız gerekir. Oksijen ekipmanınızı sağlayan
Ekipman Sağlayıcısı, reçete edilen akış hızının
nasıl ayarlanacağını gösterecektir.

Operatör ProfiliYoğunlaştırıcılar, akciğerlerin havadaki oksijeni
kan dolaşımına aktarma verimliliğini etkileyen
hastalıklardan dolayı rahatsızlık çeken kullanıcılara
ek oksijen sağlamak üzere tasarlanmıştır.
Taşınabilir oksijen yoğunlaştırıcılar (POC'ler)oksijen depolamaz veya içermez. Yeniden
doldurulmaları gerekmeyen bu yoğunlaştırıcılar,bir AC veya DC güç kaynağının mevcut olduğu
her yerde şarj edilebilir (uçak dışında). Bu dakullanıcıyı evde kullanım, (evin hem dışında hemde içinde) yürüme, hareketlilik ve genel yaşam tarzı
konusunda görece bağımsız hale getirir. Oksijenyoğunlaştırıcı, bir hekim tarafından reçete edilmesi
durumunda kullanılır.
Oksijen terapisinin uygulandığı kişiler genellikle
65 yaşın üstündedir ve bu kişiler, Kronik Obstrüktif
Akciğer Hastalığı (KOAH) dahil olmak üzere
çeşitli solunum hastalıklarından muzdariptir.
Kullanıcıların bilişsel yetileri genellikle iyi
durumdadır ve rahatsız olduklarını ifade
edebilmeleri gerekir. Kullanıcı, rahatsızlıklarını
iletebilecek durumda değilse veya yoğunlaştırıcı
etiketiyle kullanım talimatlarını okuyamayacakve anlayamayacak durumdaysa, cihazın yalnızca
bu konularda yardımcı olabilecek birinin
gözetiminde kullanılması önerilir. Yoğunlaştırıcının
kullanımı sırasında herhangi bir rahatsızlığın
hissedilmesi durumunda kullanıcıların, sağlık
hizmeti sağlayıcılarıyla iletişime geçmeleriönerilir. Kullanıcılara ayrıca, güç kesintisi veya
yoğunlaştırıcının arızalanması ihtimaline karşı
yedek oksijen kaynağı (yani silindir tüpte sağlanan
oksijen) bulundurmaları önerilir. Yoğunlaştırıcı
kullanımı için gerekli olan başka herhangi bir
beceri veya yeterlilik söz konusu değildir.

FreeStyle ComfortÖnemli!
Güvenlik Talimatları aşağıdaki gibi tanımlanmıştır:UYARI: CİDDİ YARALANMAYA NEDEN
OLABİLECEK TEHLİKELERE DAİRÖNEMLİ GÜVENLİK BİLGİLERİ.
DİKKAT: FreeStyle Comfortünitesinin zarar görmesini önlemekiçin gerekli bilgiler.
Not: Özellikle dikkat edilmesi gereken bilgiler.

Kullanım Endikasyonları
UYARI: YALNIZCA REÇETE EDILEN OKSIJEN DÜZEYININ SEÇILMESI BÜYÜK
ÖNEM TAŞIR. LISANSLI BIR KLINISYENTARAFINDAN BU YÖNDE TALIMATVERILMEDIĞI SÜRECE AKIŞ SEÇIMINIDEĞIŞTIRMEYIN.
UYARI: BU CİHAZ PEDİYATRİK HASTALARDAKULLANIM İÇİN TEST EDİLMEMİŞ VEYAONAYLANMAMIŞTIR.
UYARI: FEDERAL (ABD) YASALAR, BU CIHAZINYALNIZCA BIR HEKIM VEYA LISANSLI BAŞKA
BIR SAĞLIK HIZMETI SAĞLAYICISI TARAFINDANYA DA BU KIŞILERIN TALIMATIYLA SATILMASINIVEYA KIRALANMASINI GEREKLI KILAR.
UYARI: BU ÜNITE YAŞAM DESTEK ÜNITESIOLARAK KULLANILAMAZ. BU MAKİNEYİ
KULLANIRKEN RAHATSIZLIĞINI İLETEMEYECEKOLAN YAŞLILAR, ÇOCUKLAR VEYADİĞER KULLANICILARIN İLAVE İZLEMEYE
İHTİYAÇLARI OLABİLİR. İŞITME VE/VEYAGÖRME BOZUKLUKLARI OLAN KULLANICILAR,
ALARMLARI IZLEME KONUSUNDA YARDIMAIHTIYAÇ DUYABILIR.
UYARI: PULS DOZU MODU AYARLARI, DINLENME,EGZERSIZ VE SEYAHAT SIRASINDAKI
IHTIYAÇLARINA GÖRE HER BIR KULLANICI IÇINAYRI AYRI BELIRLENMELIDIR.

UYARI: HASTANIN SOLUNUM ÇABALARI,
KORUMA FONKSİYONUNU TETİKLEMEYEBİLİR.
UYARI: FREESTYLE COMFORT, TRAKEOSTOMİUYGULANMIŞ HASTALARDA KULLANILMAKÜZERE TASARLANMAMIŞTIR.
UYARI: HAMİLE VEYA EMZİREN KADINLAR,FTALATLAR İÇERDİĞİ İÇİN BU EL KİTABINDA
ÖNERİLEN AKSESUARLARI KULLANMAMALIDIR.

Kullanım Kontrendikasyonları
UYARI: BELİRLİ DURUMLARDA,REÇETELENMİŞ OLANIN DIŞINDA
OKSİJEN KULLANIMI TEHLİKELİ
OLABİLİR. BU CİHAZ YALNIZCA
BİR HEKİM TARAFINDAN REÇETE
EDİLDİĞİNDE KULLANILMALIDIR.
UYARI: BU CİHAZ, MRG MAKİNELERİ GİBİMANYETİK REZONANS ORTAMLARINDA
VEYA YAKININDA KULLANILMAK İÇİNTASARLANMAMIŞTIR.
UYARI: YANICI ANESTETİKLERİN BULUNDUĞU
YERLERDE KULLANILMAMALIDIR.
UYARI: ELEKTRIKLE ÇALIŞAN TÜM CIHAZLARDAOLDUĞU GIBI KULLANICI, ELEKTRIKKESINTISI NEDENIYLE KULLANIM DIŞI KALMASÜRELERIYLE KARŞI KARŞIYA KALABILIR YADA OKSIJEN YOĞUNLAŞTIRICIYA YETKILI BIRTEKNISYENIN SERVIS IŞLEMI UYGULAMASIGEREKEBILIR. OKSIJEN YOĞUNLAŞTIRICI, BU
TÜR GEÇICI KESINTILER SONUCUNDA SAĞLIK
AÇISINDAN OLUMSUZ SONUÇLARLA KARŞI
KARŞIYA KALABILECEK KULLANICILAR IÇINUYGUN DEĞILDIR.
UYARI: BU CİHAZIN TÜM ORTAMLARDAKULLANIMI ONAYLANMAMIŞTIR. BELİRLİ
ELEKTROMANYETİK KAYNAKLARA MARUZİYET
PARAZİTE VEYA İSTENMEYEN CİHAZPERFORMANSINA NEDEN OLABİLİR. PARAZİT
DURUMUNDA CİHAZ DERHAL KAPATILMALIDIR.

UYARI: BU CİHAZ EVDE SAĞLIK VE BENZERAYAKTA HASTA ORTAMLARINDA KULLANIM
İÇİN TASARLANMIŞTIR.
UYARI: BU CİHAZ OKSİJEN TEDAVİSİ ALAN
HASTALARCA KULLANILMAK ÜZERETASARLANMIŞTIR.

PN MN232-C4 L | Kullanıcı El Kitab TR - 415

FreeStyle Comfort
Güvenlik Kılavuzları
UYARI: BU EKIPMAN ÜZERINDEHERHANGI BIR DEĞIŞIKLIK YAPILMASIYASAKTIR.
UYARI: BU CIHAZ, HIZLI YANMAYIKOLAYLAŞTIRAN YÜKSEK YOĞUNLUKLUOKSIJEN SAĞLAMAKTADIR. (1) BU CIHAZIN
VEYA (2) OKSIJEN TAŞIYAN HERHANGI BIRAKSESUARIN BULUNDUĞU ODADA SIGARAIÇILMESINE VEYA AÇIK ATEŞ OLMASINA IZINVERMEYIN. BU UYARIYA UYULMAMASI CIDDI
YANGINLARA, MADDI HASARA YOL AÇABILIRVE/VEYA FIZIKSEL YARALANMA YA DA ÖLÜMESEBEBIYET VEREBILIR.
UYARI: ÜNITEYI, HAVALANDIRMANIN SINIRLIOLABILECEĞI KISITLI VEYA SIKIŞIK BIR ALANDA(ÖR. KÜÇÜK BIR KILIF VEYA EL ÇANTASIIÇINDE) ÇALIŞTIRMAYIN. BU DURUM, OKSIJEN
YOĞUNLAŞTIRICININ AŞIRI ISINMASINA VE
PERFORMANSININ DÜŞMESINE NEDEN OLABILIR.
UYARI: YOĞUNLAŞTIRICININ, DUMAN, KIRLETICIMADDE VEYA ISTEN UZAK BIR YERDEKONUMLANDIRILMASI GEREKIR.
UYARI: BU CIHAZLA ILGILI CIDDI BIR OLAY
YAŞANMASI DURUMUNDA, KULLANICI DURUMU
DERHAL SAĞLAYICIYA VE/VEYA ÜRETICIYEBILDIRMELIDIR. CIDDI OLAY; YARALANMA, ÖLÜMVEYA YENIDEN MEYDANA GELMESI DURUMUNDAYARALANMAYA/ÖLÜME NEDEN OLMA OLASILIĞI
OLARAK TANIMLANIR. KULLANICI AYRICA OLAYIMEYDANA GELDIĞI ÜLKEDEKI YETKILI MAKAMABILDIREBILIR.
UYARI: BU OKSIJEN YOĞUNLAŞTIRICIDA KULLANIM IÇIN BELIRTILMEMIŞ
OLAN OKSIJEN UYGULAMA AKSESUARLARININ KULLANILMASI, ÜNITEPERFORMANSININ DÜŞMESINE YOLAÇABILIR. ÖNERILEN AKSESUARLAR
BU EL KITABINDA BELIRTILMIŞTIR.
UYARI: OKSIJEN YOĞUNLAŞTIRICI DÜŞTÜYSE,HASAR GÖRDÜYSE YA DA SUYA MARUZKALDIYSA LÜTFEN ÜNITENIN INCELENMESI VEYAOLASI ONARIMI IÇIN EVDE BAKIM SAĞLAYICINIZ
ILE ILETIŞIME GEÇIN. GÜÇ KABLOSU YA DAFIŞI HASARLIYSA OKSIJEN YOĞUNLAŞTIRICIYIKULLANMAYIN.
UYARI: HASARI VEYA YARALANMAYI ÖNLEMEKIÇIN TAŞIMA SIRASINDA TAŞIT IÇINDE OKSIJEN
YOĞUNLAŞTIRICIYI DÜZGÜN BIR ŞEKILDESABITLEYIN, KEMERLE BAĞLAYIN VEYA BAŞKA
BIR ŞEKILDE SABIT TUTUN.
416 - TR PN MN232-C4 L | Kullanıcı El Kitab

UYARI: ABD ULAŞTIRMA BAKANLIĞI (DOT) VE
BIRLEŞMIŞ MILLETLER (BM) DÜZENLEMELERI,OKSIJEN YOĞUNLAŞTIRICININ BAGAJIÇINDE HAVAYOLUYLA TAŞINDIĞI TÜM
ULUSLARARASI SEYAHATLERDE PILIN CIHAZDANÇIKARILMASINI GEREKLI KILMAKTADIR. OKSIJEN
YOĞUNLAŞTIRICININ NAKLIYATI SIRASINDADA PIL CIHAZDAN ÇIKARILMALI VE DÜZGÜN
ŞEKILDE PAKETLENMIŞ OLMALIDIR.
UYARI: BU EKİPMANI, ÖNCELİKLE BU EL KİTABINIOKUYUP ANLADIĞINIZDAN EMİN OLDUKTANSONRA ÇALIŞTIRIN. UYARI VE TALİMATLARI
ANLAMAKTA SIKINTI YAŞIYORSANIZ BUEKİPMANI KULLANMADAN ÖNCE EKİPMAN
SAĞLAYICINIZLA İLETİŞİME GEÇİN; AKSİ HALDEYARALANMA VEYA HASAR MEYDANA GELEBİLİR.
UYARI: HERHANGI BIR RAHATSIZLIĞINIZ YA DAACIL ILGILENILMESI GEREKEN BIR HASTALIĞINIZVARSA VAKIT KAYBETMEDEN TIBBI DESTEK IÇINBAŞVURUN.
UYARI: OKSIJEN YOĞUNLAŞTIRICININ,ÇALIŞTIRMA SICAKLIĞI TEKNIK ÖZELLIKLERINDEBELIRTILENIN DIŞINDA BIR SICAKLIKTA
ÇALIŞTIRILMASI, YOĞUNLAŞTIRICININ OKSIJEN
YOĞUNLUĞU TEKNIK ÖZELLIKLERINE UYGUN
ÇALIŞMA BECERISINI SINIRLAYABILIR. SICAKLIKSINIRLARI IÇIN BU EL KITABININ TEKNIKÖZELLIKLER BÖLÜMÜNE BAKIN.
UYARI: OKSIJEN TAŞINAN AKSESUARLAR VEYAOKSIJEN YOĞUNLAŞTIRICI ILE YAĞ, GRES YA DAPETROL BAZLI VEYA DIĞER YANICI ÜRÜNLERIKULLANMAYIN. YALNIZCA SU BAZLI, OKSIJENILE UYUMLU LOSYON YA DA MERHEMLERKULLANILMALIDIR. OKSIJEN, YANICIMADDELERIN TUTUŞMASINI HIZLANDIRIR.
UYARI: PILIN HATALI KULLANILMASI, PILIN
ISINMASINA VE TUTUŞMASINA YOL AÇARAK
CIDDI YARALANMALARA NEDEN OLABILIR. PILIKESINLIKLE DELMEYIN, DÜŞÜRMEYIN, PILE
VURMAYIN, PILIN ÜZERINE BASMAYIN VEYA PILIBAŞKA BIR ŞEKILDE SERT DARBELERE YA DAŞOKLARA MARUZ BIRAKMAYIN. HASARLI PILKULLANIMI, FIZIKSEL YARALANMAYA NEDEN
OLABILIR.
UYARI: PILIN METAL TEMAS UÇLARINI ANAHTAR
VEYA BOZUK PARA GIBI METALIK NESNELERLEKISA DEVRE YAPTIRMAYIN. BU, KIVILCIM VEYAAŞIRI ISI OLUŞUMUNA NEDEN OLABILIR.

FreeStyle Comfort
UYARI: TAŞINABILIR OKSIJEN YOĞUNLAŞTIRICIYIAC GÜÇ KAYNAĞI ILE DIŞ MEKANLARDAKULLANIRKEN, GÜÇ KAYNAĞINI YALNIZCATOPRAKLAMA KAÇAK DEVRE KESICISI (GFI)
BULUNAN BIR PRIZE BAĞLAYIN.
UYARI: OKSIJEN YOĞUNLAŞTIRICI, BAŞKA
EKIPMANLARLA BITIŞIK VEYA ÜST ÜSTEKONMUŞ OLARAK KULLANILMAMALIDIR.
BAŞKA EKIPMANLARLA BITIŞIK YA DA ÜST ÜSTEKONMUŞ OLARAK KULLANILMASI GEREKIYORSACIHAZIN NORMAL ŞEKILDE ÇALIŞIP ÇALIŞMADIĞIGÖZLEMLENMELIDIR.
UYARI: TIBBI ELEKTRIKLI EKIPMANIN ÜRETICISITARAFINDAN IÇ BILEŞENLERIN YEDEKPARÇALARI OLARAK SATILAN KABLOLAR
VE ADAPTÖRLER HARIÇ OLMAK ÜZERE,BELIRTILENLERIN DIŞINDAKI KABLO VE
ADAPTÖRLERIN KULLANILMASI, OKSIJEN
YOĞUNLAŞTIRICININ BAĞIŞIKLIK SISTEMININ
ZAYIFLAYARAK DAHA FAZLA EMISYONSALMASINA SEBEBIYET VEREBILIR.
UYARI: YALNIZCA CIHAZA YAPIŞTIRILMIŞSPESIFIKASYON ETIKETI ÜZERINDE BELIRTILENELEKTRIK VOLTAJINI KULLANIN.
UYARI: BU ÜNITEYI UZATMA KABLOSUYLAKULLANMAYIN VEYA AYNI ELEKTRIK PRIZINE ÇOK
SAYIDA FIŞ TAKMAYIN. UZATMA KABLOSUYLAKULLANILMASI CIHAZIN PERFORMANSINIOLUMSUZ YÖNDE ETKILEYEBILIR. TEK BIR
PRIZE ÇOK SAYIDA FIŞ TAKILMASI, ELEKTRIKPANELINDE AŞIRI YÜKE NEDEN OLARAK DEVREKESICININ/SIGORTANIN DEVREYE GIRMESINE YA
DA DEVRE KESICI VEYA SIGORTA ÇALIŞMAZSAYANGINA SEBEP OLABILIR.
DİKKAT: Alarm durumunda veya
oksijen yoğunlaştırıcının doğru şekildeçalışmadığını gözlemlediğinizde, bu elkitabındaki Sorun Giderme bölümüne
bakın. Sorunu çözemezseniz EkipmanSağlayıcınıza danışın.

UYARI: BU EL KITABINDA BELIRTILEN
OLASI ÇÖZÜMLER DIŞINDA HERHANGIBIR BAKIM IŞLEMINI DENEMEYIN.
UYARI: SADECE ÜRETICI TARAFINDAN TAVSIYE
EDILEN AKSESUARLARI KULLANIN. BAŞKAHERHANGI BIR AKSESUARIN KULLANILMASIOKSIJEN YOĞUNLAŞTIRICINIZ IÇIN ZARARLIOLABILIR, CIDDI HASARA YOL AÇABILIR VEGARANTIYI GEÇERSIZ KILAR.
UYARI: ÜRETICI, ELEKTRIK KESINTISI, ALARM
DURUMU VEYA MEKANIK ARIZA IHTIMALINEKARŞI ALTERNATIF BIR TAMAMLAYICI OKSIJEN
KAYNAĞININ HAZIR BULUNDURULMASINIÖNERIR. GEREKLI OLAN YEDEK SISTEMINTÜRÜNÜ ÖĞRENMEK IÇIN HEKIMINIZE YA DAEKIPMAN SAĞLAYICINIZA DANIŞIN.
UYARI: HAVA GIRIŞI YA DA HAVA ÇIKIŞIDELIKLERININ ENGELLENMESINE IZIN VERMEYIN.
CIHAZ ÜZERINDEKI AÇIKLIKLARDAN HIÇBIRININIÇINE NESNE DÜŞÜRMEYİN VEYA SOKMAYIN.
BU DURUM, OKSIJEN YOĞUNLAŞTIRICININ AŞIRI
ISINMASINA VE PERFORMANSININ DÜŞMESINENEDEN OLABILIR.
UYARI: OKSIJEN KULLANIRKEN SIGARA IÇMEK,YANGINDAN KAYNAKLANAN YARALANMA VE
BUNA BAĞLI ÖLÜMLERIN EN YAYGIN NEDENIDIR.
ŞU GÜVENLIK UYARILARINI IZLEMELISINIZ:UYARI: CIHAZIN YA DA OKSIJEN TAŞINANAKSESUARLARIN BULUNDUĞU ODADASIGARA IÇILMESINE, MUM VEYA AÇIK ALEVBULUNMASINA IZIN VERMEYIN.
UYARI: OKSIJEN KANÜLÜ TAKILIYKEN SIGARAIÇMEK, YÜZDE YANIKLARA NEDEN OLARAKÖLÜMLE SONUÇLANABILIR.
UYARI: SIGARA IÇIYORSANIZ BU 3 ADIM
HAYATINIZI KURTARABILIR: OKSIJEN
YOĞUNLAŞTIRICIYI KAPATIN, KANÜLÜ ÇIKARINVE CIHAZIN BULUNDUĞU ODADAN ÇIKIN.

DİKKAT: Garantinin geçersiz hale gelmesiniönlemek için üreticinin tüm talimatlarını izleyin.

PN MN232-C4 L | Kullanıcı El Kitab TR - 417

FreeStyle Comfort
UYARI: EVDE VEYA OKSIJENIN KULLANILDIĞIHER YERDE HER ZAMAN "SIGARA İÇILMEZ OKSIJEN KAYNAĞI" BENZERI LEVHALARBULUNMALIDIR. KULLANICILAR VE BAKICILARI,
TIBBI OKSIJENIN BULUNDUĞU ORTAMDAVEYA KULLANILMASI SIRASINDA SIGARAIÇMENIN TEHLIKELERI HAKKINDA BILGI SAHIBIOLMALIDIR.
UYARI: OKSIJEN YOĞUNLAŞTIRICINIZI YANICIGAZLARIN BULUNDUĞU ORTAMDA KULLANMAYIN.
BU DURUM HIZLA YANMAYA YOL AÇARAK MADDI
HASARA, VÜCUTTA YARALANMALARA VEYAÖLÜME SEBEBIYET VEREBILIR.
UYARI: KANÜLÜN ÇIKARILMASI VE GIYSILERIN,YATAĞIN, KANEPENIN YA DA YASTIK BENZERIDIĞER EŞYALARIN ÜZERINE KONULMASI,SIGARA, ISI KAYNAĞI VEYA ALEVE MARUZKALDIĞINDA PARLAYARAK ALEV ALMASINA
NEDEN OLACAKTIR.
UYARI: BURUN KANÜLÜNÜ GIYSILERIN, YATAKÖRTÜLERININ VEYA SANDALYE MINDERLERININÜZERINDE YA DA ALTINDA BIRAKMAYIN. ÜNITE
AÇIK OLDUĞU HALDE KULLANIMDA DEĞILSEOKSIJEN BU EŞYALARIN YANICI OLMASINASEBEBIYET VERIR. OKSIJEN YOĞUNLAŞTIRICININ
KULLANILMADIĞI SIRADA GÜCÜNÜN KAPALIKONUMDA OLDUĞUNDAN EMIN OLUN.
UYARI: ONAYLI OKSIJEN DAĞITIMAKSESUARLARI, HASTANIN GÜVENLIĞI
AÇISINDAN YANGININ YAYILMASINI AZALTACAKŞEKILDE TASARLANMIŞTIR. ONAYLANMAYAN
AKSESUARLARIN KULLANILMASI YARALANMAVEYA YANGINA BAĞLI HASAR RISKINI ARTIRABILIR.
UYARI: PİL TAKIMI ÇIKARILMADAN ÖNCE,OKSİJEN YOĞUNLAŞTIRICI KAPALI KONUMDA
OLMALIDIR VE HARİCİ BİR GÜÇ KAYNAĞINA
BAĞLI OLMAMALIDIR.
UYARI: KÜÇÜK PARÇALAR, BOĞULMATEHLIKESINE NEDEN OLABILIR.
UYARI: SICAK OLABILECEĞI IÇIN EGZOZ HAVA
AĞZIYLA 10 SANIYEDEN UZUN SÜRE TEMASETMEKTEN KAÇININ.
UYARI: RÜZGAR VEYA GÜÇLÜ HAVA AKIMLARI,OKSIJEN TEDAVISININ DOĞRU ŞEKILDEUYGULANMASINI OLUMSUZ ETKILEYEBILIR.
UYARI: BU ÜRÜN, CALIFORNIA EYALETITARAFINDAN KANSERE NEDEN OLDUĞU BILINEN
NIKEL GIBI KIMYASAL MADDELERE MARUZ
KALMANIZA YOL AÇABILIR. DAHA FAZLA BILGIIÇIN WWW.P65WARNINGS.CA.GOV WEB SITESINIZIYARET EDIN.
418 - TR PN MN232-C4 L | Kullanıcı El Kitab

DİKKAT: AC güç kaynağını oksijenyoğunlaştırıcıdan ayırmadan önce her
zaman AC güç kaynağını prizden çekin.
DİKKAT: Oksijen yoğunlaştırıcıyı, güç kablosunaerişimi güçleştirecek şekilde konumlandırmayın.
DİKKAT: Oksijen kaynağı borularını ve güç
kablolarını, insanların takılmasını veya olası birdolanmayı engelleyecek biçimde yerleştirin.
DİKKAT: Oksijen yoğunlaştırıcıyı otomobilde, teknede
veya DC güç kaynağına sahip diğer DC kaynaklarıüzerinde kullanırken, ürünü bağlamadan önce taşıtın
çalıştırıldığından ve çalışır durumda olduğundanemin olun. Bu talimatların uygulanmaması,
güç kaynağının oksijen yoğunlaştırıcıya güçsağlamamasıyla sonuçlanabilir.
DİKKAT: Oksijen Yoğunlaştırıcı ünitesini kullandığınız
otomobil çalışmaz durumdayken, cihazın otomobilleolan bağlantısını kesin ve cihazı çıkarın. OksijenYoğunlaştırıcıyı çok sıcak veya soğuk bir otomobilde
veya yüksek ya da düşük sıcaklığa sahip diğer
benzer ortamlarda saklamayın.
Not: Taşınabilir ve mobil RF haberleşme ekipmanı
tıbbi elektrikli ekipmanı etkileyebilir.
DİKKAT: Marş KAPALI konumdayken,
oksijen yoğunlaştırıcıyı veya güç
kaynağını taşıta takılı şekilde
BIRAKMAYIN. Bu, taşıtın pilini tüketebilir.
DİKKAT: Oksijen yoğunlaştırıcının normal çalışma
sıcaklığı aralığı ve ortam basıncı aralığı (irtifa)dışında çalıştırılması veya saklanması, performansı
etkileyebilir ve pil çalışma süresini azaltabilirve/veya pil şarj süresini artırabilir. Saklama veçalışma sıcaklığı sınırları için bu el kitabının
Spesifikasyonlar bölümüne başvurun.
DİKKAT: Pil donanımlı oksijen yoğunlaştırıcılariçin: Pilinizin uzun ömürlü olmasını sağlamak içinpilleri serin ve kuru bir konumda saklayın. Oksijenyoğunlaştırıcınızın uzun bir süre boyunca yüksek
sıcaklıklarda veya tam şarj edilmiş ya da tamamenboşalmış bir pille saklanması, genel pil ömrünü
azaltabilir. Pilin içini açma girişiminde bulunmayın;
pil içinde servis işlemi uygulanabilen parça yoktur.
Pilleri çocukların erişemeyeceği konumda tutun.
DİKKAT: YALNIZCA Üretici tarafından sağlananpilleri KULLANIN. Pillerin doğru şekilde eldençıkarılması için imha gereksinimleriyle ilgili olarak
Ekipman Sağlayıcınız veya yerel idari birimle
iletişime geçin.

FreeStyle Comfort
Not: Yoğunlaştırıcının veya elektrik kesintisinin
neden olduğu zararı azaltmak için, ekipmansağlayıcısı kurulum sırasında ve bu özelliklerdeğiştiğinde periyodik olarak yedek oksijen ihtiyacınıdeğerlendirmelidir:

Not: Oksijen yoğunlaştırıcı spesifikasyon tablosunda
belirtilen normal çevresel çalışma koşullarınındışında uzun bir süre boyunca saklanmışsa üniteninçalıştırılmadan önce normal çalışma sıcaklığına geri
dönmesine izin verilmelidir.

- Hastanın durumu

Not: Ekipman sağlayıcısı kurulum sırasındayoğunlaştırıcı veya güç arızası durumunda ve buözellikler değiştiğinde periyodik olarak yedek oksijen
ihtiyacını değerlendirmelidir:- Hastanın yaşadığı ortam- Hastaya yedek oksijen desteği sağlama yeteneği.
Not: Kanülün tamamen takılı olduğundan vesabitlendiğinden emin olun. Oksijen yoğunlaştırıcı, bu
sayede oksijen uygulaması için nefes almayı doğru
şekilde algılar. Nefes alma sırasında burun kanülünün
sivri uçlarına oksijen akışı olduğunu duymanız ya dahissetmeniz gerekir.

- Hastanın durumu- Hastanın yaşadığı ortam- Hastaya yedek oksijen desteği sağlama yeteneği.

Not: Lityum piller, tam şarj edilmiş veya tamamenboşaltılmış durumdayken aşırı yüksek sıcaklıklara
maruz bırakıldığında kalıcı kapasite kaybına uğrar.
Uzun süreli saklama için pillerin %25-50 oranında şarjedilmesi ve -20°C ila +20°C (-4°F ila 68°F) sıcaklık
aralığında tutulması önerilir.
Not: Oksijen yoğunlaştırıcıyı, hava girişi filtresi takılı
olmadan çalıştırmayın. İkinci bir filtre sağlanmışsa
kirli filtreyi temizlemeden önce yedek filtreyi takın.
Kirli filtreyi ılık su ve sabun çözeltisinde temizleyin,
ardından kullanmadan önce tamamen kurutun.
Not: Üretici bu ekipmanın sterilize edilmesiniönermemektedir.
Not: Oksijen yoğunlaştırıcı ünitenizi kullanırken
odadaki oksijenin tükenme tehlikesi asla söz konusudeğildir.
UYARI: OKSIJEN YOĞUNLAŞTIRICI,IÇERISINDE PIL OLMADAN
ÇALIŞTIRILMAMALI VE HARICIBIR (AC/DC) GÜÇ KAYNAĞINA
BAĞLANMAMALIDIR. PIL TAKIMIÇIKARILMIŞSA ÜNITEYI ÇALIŞTIRMAYIVEYA HARICI BIR GÜÇ KAYNAĞINA
BAĞLAMAYI DENEMEYIN.
PIL TAKIMINI ANCAK ÜNITE KAPALI KONUMAGETIRILDIKTEN VE ÜNITENIN GÜÇ KAYNAĞI
BAĞLANTISI KESILDIKTEN SONRA ÇIKARIN.
ÜNITE AÇIK KONUMDA VEYA GÜÇ KAYNAĞINA
BAĞLI DURUMDAYKEN PIL TAKIMINI ÇIKARMAYIN.

PN MN232-C4 L | Kullanıcı El Kitab TR - 419

FreeStyle Comfort
Teknik Özellikler
FreeStyle Comfort
Boyutlar (Tek Pil ile)

10,0 x 7,3 x 3,1 inç (25,4 x 18,5 x 7,9 cm)

Boyutlar (Çift Pil ile)

11,0 x 7,3 x 3,1 inç (27,9 x 18,5 x 7,9 cm)

Ağırlık (Tek Pil Takımı ile)

2,3 kg (5 lb)Ağırlık (Çift Pil Takımı ile)

2,7 kg (6 lb)

Anma Ses Düzeyi***39,93 dB(A) ± 0,42 dB(A) @ 2Akış Ayarları Puls Ayarı 1, 2, 3, 4, 5
O2 Çıkış Ayarı Ayar 1 - 210 mL/dak
Ayar 2 - 420 mL/dak
Ayar 3 - 630 mL/dak
Ayar 4 - 840 mL/dak
Ayar 5 - 1.050 mL/dk([@ 20 bpm STPD] Tüm Hacimler +/- %15)

Oksijen Yoğunluğu*Oksijen Monitörü
Maksimum Çıkış Basıncı
Dozlama Hassasiyeti%90 (+%5,5/-%3)Evet<30 PSI>-0,1 cm H2O (ayarlanabilir)

AC Güç 100-240 VAC, 50-60 Hz, 120 W Maks
DC Güç

11-18 VDC (10 maks amp)

Pil Türü

Lityum İyonPil Kapasitesi

Tek hücre: 6.700 mAh; Çift hücre: 13.400 mAh

Pil Şarj Süresi

Tek Pil: 3,5 saatÇift Pil: 6 saat

Pil Takımı Süreleri

Tek Pil Süresi: Ayar 2 - en fazla 4 saatÇift Pil Süresi: Ayar 2 - en fazla 8 saatÇalışma Sıcaklığı**5°C - 40°C (41°F - 104°F)Taşıma ve Depolama Sıc. (Ünite)

3.048 m’ye kadar (10.000 fit) -25°C ila +70°C (-13°F ila +158°F)Taşıma ve Depolama Sıc. (Pil)3.048 m’ye kadar (10.000 fit) -25°C ila +70°C (-13°F ila +158°F)Çalışma İrtifası** -381 ila 3.048 m (-1.250 ila 10.000 fit) (700 - 1.060 hPa’da
test edilmiştir)Çalışma Nemliliği

%15 - 95 bağıl nem (yoğuşmasız)

Depolama Nemi

%0 - 90 Yoğuşmasız

Sınırlı Garanti

Garanti Beyanına Bakın* 21°C'de (70°F) 14,7 psi (101 kPa) atmosfer basıncı temel alınmıştır.
** Ortam belirtilen sıcaklık ve nem aralıklarında olduğu sürece ünite tüm ortamlarda işlev gösterecektir. Bu kullanımkoşulları dışında çalıştırmak, yoğunlaştırıcının daha yüksek litreli akış hızlarında Oksijen Yoğunlaştırma koşullarınıkarşılama becerisini sınırlayabilir. Isınma ekranı, yoğunlaştırıcı hedef yoğunluğa ulaşana dek veya en fazla 15 dakika
boyunca görünmeye devam edebilir. Bu süre boyunca üniteyi kullanmaya devam edin.
*** Test yöntemi No. 14-1 10/2018 MDS-Hi uyarınca ölçülen ses seviyesi.
Öngörülen kullanım ömrü en az beş yıldır.

420 - TR PN MN232-C4 L | Kullanıcı El Kitab

FreeStyle ComfortÇalıştırma Talimatları
1. FreeStyle Comfort cihazını, kullanım için uygun
bir konuma yerleştirin.
a. H
 ava girişi filtrelerinin ve hava çıkış kanalının
tıkanmaması için yoğunlaştırıcının yeterli bir
biçimde havalandırılması gereklidir.
b. Yoğunlaştırıcının, her türlü işitsel/görsel
gösterge ve alarmın kolayca görülebilecekve işitilebilecek biçimde konumlandırılması
gereklidir.
c. Üniteyi çalıştırmadan önce hava girişi
filtrelerinin yerinde olduğundan eminolun. Ünitede bir filtre yoksa ekipmansağlayıcınızla iletişime geçin.
2. Yoğunlaştırıcıya bir Pil Takımı takın.
a. Pili, FreeStyle Comfort’ın merkezindeki alt
muhafazayla hizalayın.

3. AC veya DC güç kaynağını, ünitenin ön sağ alt
tarafında bulunan güç portu prizine takın.

4. Burun kanülünü oksijen çıkış ucuna bağlayın.
Cihazı çalıştırmadan önce bağlantınıngerçekleştiğinden emin olun.
NOT: Önerilenler dışındaki kanüllerin kullanılması
performansı etkileyebilir.

Not: Ekipman Sağlayıcısına: FreeStyle Comfort
ile aşağıdaki oksijen uygulama aksesuarlarının
kullanılması önerilir:b. Pilin yerine oturup kilitlendiğini belirten bir
tıklama sesi duyana kadar pili içeri doğru
kaydırın.

• 2,1 m (7 fit) hortum (Maksimum LPM: 6 LPM) ile birlikte
Burun Kanülü CAIRE Parça Numarası 5408-SEQ• Önerilen ek aksesuar bilgileri www.caireinc.com
adresinde çevrimiçi olarak mevcuttur.
5. Güç düğmesine basarak FreeStyle Comfort’ı
açın ve akışı artırma (+) veya azaltma (-)
düğmelerini kullanarak reçete edilmiş olan akışayarınızı seçin.

PN MN232-C4 L | Kullanıcı El Kitab TR - 421

FreeStyle Comfort
Nefes
Hızı
15
20
25
30
35
40
Cihaza güç verildiğini belirtmek için kısa bir sesli
alarm duyulacaktır.
Not: FreeStyle Comfort açıldığında, varsayılanakış ayarı 2’dir. Ayar değiştirilirse ünite önceki akışayarında açılacaktır.
6. Burun kanülünü, kanülle birlikte verilen üretici
talimatlarına uygun olarak yüzünüze yerleştirin
ve normal şekilde nefes alıp vermeye başlayın.
Her soluk aldığınızda, burun kanülünden verilen
bir oksijen akışı hissetmeniz gerekir.
7. Akış ayarını değiştirmek için akışı artırma (+)
veya azaltma (-) düğmeleriyle istenen akış hızını
ayarlayın. Akış ayarlarını değiştirdiğinizde farklı
bir ses duymanız normaldir.

autoSat olmayan Puls Hacmi (mL)* Ayarı
1
2
3
4
5
14,0
10,5
8,4
7,0
6,0
5,3

28,0
21,0
16,8
14,0
12,0
10,5

42,0
31,5
25,2
21,0
18,0
15,8

56,0
42,0
33,6
28,0
24,0
21,0

70,0
52,5
42,0
35,0
30,0
26,3

8. Güç düğmesine basarak FreeStyle Comfort’ı
kapatın.
Not: Görüntü ekranı, bekleme moduna girecek
ve 20 saniye sonra kararacaktır. Ekranı bekleme
modundan çıkarmak için herhangi bir düğmeye basın.
9. Gelişmiş Özellikler için Menü Gezinme
Talimatlarına bakın.

ÖZEL ÜRÜN ÖZELLIKLERI
autoSAT®
CAIRE'nin tescilli autoSAT özelliği, FreeStyle
Comfort'un maksimum nefes hızına kadar nefesbaşına tutarlı miktarda oksijen (bolus) iletmesine
olanak tanır. Yürüme gibi aktiviteler sırasında
kullanıcının nefes hızı arttıkça, autoSAT özelliği
cihazın çalışmasını ayarlanan bolus hacmini
koruyacak şekilde ayarlar.
UltraSense®
Bu hassas puls dozu tetikleyicisi, solunumunuz
sırasındaki basınç değişimini algılar ve siz nefes
alırken oksijen pulsu verir.

Nefes
Hızı
15
20
25
30
35
40

autoSat Puls Hacmi (mL)* Ayarı
1
2
3
4

5

10,5
10,5
10,5
10,5
10,5
10,5

52,5
52,5
42,0
35,0
30,0
26,3

21,0
21,0
21,0
21,0
21,0
21,0

31,5
31,5
31,5
31,5
30,0
26,3

42,0
42,0
42,0
35,0
30,0
26,3

*Tolerans +/- %15’tir

422 - TR PN MN232-C4 L | Kullanıcı El Kitab

autoDOSE®
"Solunum yok alarmı" verildiğinde bu özellik,
alarm durumu giderilene kadar ünitenin mevcutakış ayarıyla dakikada 20 nefeslik oksijen sağlar.

FreeStyle Comfort
Güç Kaynakları
FreeStyle Comfort’a üç şekilde güç verilebilir:• Şarj Edilebilir Pil Takımı• AC Güç• DC Güç
Bu bileşenlerin her birinin çalıştırılmasına ilişkin
bilgiler, aşağıdaki bölümlerde verilmiştir.
UYARI: YALNIZCA BU KULLANICI EL
KITABINDA BELIRTILEN GÜÇ KAYNAKLARINI KULLANIN. CAIRE TARAFINDANÜRETILMEMIŞ GÜÇ KAYNAKLARI VE
PIL TAKIMLARININ KULLANILMASI, ÜNITEYE ZARAR VEREBILIR VE ÜNITENINDOĞRU BIR BIÇIMDE ÇALIŞMAMASINAYOL AÇABILIR.

PİLİN ÇIKARILMASI

Pil çıkarma düğmesini FreeStyle Comfort cihazından
aşağı doğru itin. Bu düğmeyi TUTARKEN, piliyoğunlaştırıcıdan kaydırarak çekip çıkarın.
UYARI: PIL TAKIMI ÇIKARILMADAN
ÖNCE, OKSIJEN YOĞUNLAŞTIRICIKAPALI KONUMDA OLMALIDIR VE
HARICI BIR GÜÇ KAYNAĞINA BAĞLI
OLMAMALIDIR.

PİL TAKIMIFreeStyle Comfort, cihazla birlikte verilen şarjedilebilir pil takımıyla çalıştırılabilir. Pil takımıylaçalıştırılırken, yoğunlaştırıcının çalıştırılmasınailişkin herhangi bir sınırlama yoktur. Pilleçalıştırılırken tüm akış hızları kullanılabilir.

PİL TAKIMININ DOLULUK DÜZEYİNİN
KONTROL EDİLMESİ
PİLİN TAKILMASI

Pili, FreeStyle Comfort’ın merkezindeki alt
muhafazayla hizalayın. Pilin yerine oturup
kilitlendiğini belirten bir tık sesi duyana kadar pili
içeri doğru itin.

Pilin FreeStyle Comfort cihazına takılıyken doluluk
düzeyini kontrol etmek için pil simgesine veekranda gösterilen kalan tahmini süreye bakın.
%80-100%60-80%40-60%20-40%20

PN MN232-C4 L | Kullanıcı El Kitab TR - 423

FreeStyle Comfort
Not: Güç kaynağı takılı ise kalan şarj yüzdesi yerinefiş göstergesi gösterilecektir.
Pilin FreeStyle Comfort cihazına takılı olmadığı
durumda doluluk düzeyini kontrol etmek içinpili yere serin ve TEST düğmesine basın. TESTdüğmesinin sağındaki pil durumu gösterge ışıkları(%25-100) yanarak pil takımının doluluk düzeyini
gösterecektir.

PİL GÜCÜYLE ÇALIŞTIRMA SÜRESİ

Harici AC veya DC güç bağlantısı kesildiğinde,şarj edilmiş bir pil takılmışsa yoğunlaştırıcı
otomatik olarak pil gücüne geçer. Yeni bir pil
takımının tipik kullanım süreleri aşağıdaki tabloda
gösterilmektedir. Solunum hızınıza bakılmaksızın,pil takımından çalışma süresi sabittir.
Puls
Ayarı

Tek (8 Hücreli)
Pil Süresi

Çift (16 Hücreli)
Pil Süresi

1

en fazla 8 saat

en fazla 16 saat2

en fazla 4 saat

en fazla 8 saat3

en fazla 3 saat

en fazla 6 saat

4

en fazla 2,25 saat

en fazla 4,5 saat

5

en fazla 2 saat

en fazla 4 saatTİPİK PİL ŞARJ ETME SÜRESİ

İLK KEZ ŞARJ ETMEFreeStyle Comfort cihazınızla birlikte verilen yeni
pil, fabrikadan çıktığında kısmen şarj edilmiştir.
İlk kullanımdan önce, AC veya DC güç kullanarak
pil takımını tamamen şarj etmeniz gerekir.

GENEL ŞARJ ETME BİLGİLERİ

Pillerin yoğunlaştırıcıya takılı, yoğunlaştırıcınında harici bir AC veya DC güç kaynağına bağlı
olduğu her zaman, FreeStyle Comfort pil takımını
dolduracaktır (uçak dışında).
Pil, %100 gösterildiğinde tamamen şarj olmuştur.
Bunun yerine pil takımı, opsiyonel masaüstü şarjcihazı kullanılarak da şarj edilebilir.

424 - TR PN MN232-C4 L | Kullanıcı El Kitab

Pil takımınızı tam şarjı bir pilden tamamen şarjetmek için gereken tipik süre, tek pil için 3,5 saatve çift pil için 6 saattir.
Not: Pil, pil paketi tamamen boşalmışsa başlangıçtaşarj etmeye başlamayabilir.

PİL HİZMET ÖMRÜFreeStyle Comfort pilleri, 300 şarj/boşalma
döngüsü kadar dayanacak şekilde tasarlanmıştır.

FreeStyle Comfort
AC ve DC Güç Kaynağı
FreeStyle Comfort, standart olarak AC ve DC
güç kaynaklarıyla birlikte sunulur.
AC güç kaynağı şunları içerir:	a. 100-240 volt, 50-60 Hz arasında çalışan
AC Güç adaptörü.
	b. Güç adaptörü ve ilgili AC priziyle bağlantı
kurmak için kullanılan AC Güç kablosu.

DC güç kaynağı, doğrudan üniteye ve DC prizine
takılan bir kablodan oluşmaktadır.

2. AC güç kablosunu AC güç adaptörüne bağlayın.

AC GÜÇ

AC güç, evde veya standart AC prizlerinin mevcutolduğu her yerde kullanılabilir. FreeStyle Comfort,
AC güçle eksiksiz çalışır. FreeStyle Comfort,
AC gücün verildiği her zaman pil takımını da şarjedecektir (uçak dışında). FreeStyle Comfort'ınçalışır durumda olup olmamasına bakılmaksızın,pil takımı şarj edilir.
UYARI: OKSIJEN YOĞUNLAŞTIRICIYAPIL TAKIMININ TAKILMADIĞI
DURUMLARDA ÜNITEYI HARICI BIR
GÜÇ KAYNAĞINA BAĞLAMAYIN.
FreeStyle Comfort ünitesini AC Güç kaynağına
bağlamak için:
1. AC güç adaptörünü, FreeStyle Comfort’ınsağ alt köşesinden ön taraftaki güç konnektörügirişine bağlayın. Bağlantının gerçekleşmesi için
içeri doğru itin.

3. AC güç kablosunun diğer ucunu prize veya
uygun olan başka bir elektrik çıkışına bağlayın.

PN MN232-C4 L | Kullanıcı El Kitab TR - 425

FreeStyle Comfort
4. Güç düğmesine basarak FreeStyle Comfort’ı
açın ve akışı artırma (+) veya azaltma (-)
düğmelerini kullanarak reçete edilmiş olan akışayarınızı seçin.

4. Güç düğmesine basarak FreeStyle Comfort’ı
açın ve akışı artırma (+) veya azaltma (-)
düğmelerini kullanarak reçete edilmiş olan akışayarınızı seçin.
Not: Pil takımının şarj edilmesi gerekirse ekrandaki
pil simgesinde tek bir turuncu çubuk olacaktır.

DC GÜÇ

DİKKAT: Olası yangını veya aşırı ısınmayı
önlemek için, 12 V DC güç çıkışında kir
bulunmadığından ve DC Güç Kaynağının
adaptör fişinin düzgün bir şekilde
takıldığından ve güvenli bir şekilde
yerinde kaldığından emin olun.

DC güç, otomobil, tekne veya diğer motorlutaşıtların aksesuar çıkışlarından kullanım
içindir. FreeStyle Comfort, DC güçle eksiksizşekilde kullanılabilir. FreeStyle Comfort, DC
gücün verildiği her zaman pil takımını da şarjeder. Pil takımı, ünite açık olduğunda da kapalı
olduğunda da şarj olur.

DİKKAT: DC güç kaynağının metal yüzeyleri
kullanım sırasında yüksek akım nedeniyle ÇOK
SICAK hale gelebilir. Dokunmadan önce yüzeylerinsoğuması için yeterli süre tanıyın.

UYARI: OKSİJEN YOĞUNLAŞTIRICIYAPİL TAKIMININ TAKILMADIĞI
DURUMLARDA ÜNİTEYİ HARİCİ BİR
GÜÇ KAYNAĞINA BAĞLAMAYIN.

DİKKAT: DC güç kaynağını herhangi bir uzatma,ayırıcı veya adaptörle kullanmayın. Onaylanmayan
aksesuarların kullanılması bunların aşırı ısınmasına
veya hasar görmesine neden olabilir.

1. Motorlu taşıtınızı çalıştırın.
2. DC güç kaynağını, FreeStyle Comfort’ın sağ alt
tarafından ön kısımdaki güç portu prizine bağlayın.
Bağlantının gerçekleşmesi için içeri doğru itin.

3. DC güç kablosunun diğer ucunu otomobile veya
başka bir DC aksesuar çıkışına bağlayın.

426 - TR PN MN232-C4 L | Kullanıcı El Kitab

DİKKAT: Cihazın 12 V DC elektrik prizine takılması,cihaz kapalı olsa bile motor çalışmıyorsa aracın
aküsünün bitmesine neden olabilir.
DİKKAT: DC güç kaynağı bağlıyken aracı takviyeyleçalıştırmayın.

FreeStyle Comfort
Temizlik, Özen ve Düzenli Bakım
RUTİN BAKIMÖnleyici Bakım
Cihaz bakım gerektirmez. Cihazın içindeki
filtreler cihazın ömrü boyunca dayanacak şekildetasarlanmıştır. Cihaz düzgün çalışmıyorsa birim
onarım için sağlayıcıya iade edilmelidir. FreeStyle
Comfort kabininin içindeki dahili bileşenlerin
bakımı, oksijen sağlayıcısı tarafından yapılmalıdır.
Ünitenin operatörü, şu bakım prosedürlerinden
sorumludur.
• Haftalık – Hava Girişi Filtrelerinin Temizlenmesi
UYARI: OKSIJEN YOĞUNLAŞTIRICININ
PLASTIK KAPAĞINI AÇMAYA VEYAÇIKARMAYA ÇALIŞMAYIN. ÜNITEYALNIZCA GEREKLI YETKINLIĞE SAHIPBIR SERVIS TEKNISYENI TARAFINDANAÇILMALIDIR.

HAVA GİRİŞİ FİLTRELERİ

Hava, oksijen yoğunlaştırıcının alt tarafından,ön kısımdaki hava girişi filtreleriyle FreeStyle
Comfort’a girer. Bu filtreler, tozun ve havadakidiğer büyük partiküllerin üniteye girişini engeller.
Hava girişi filtrelerinin haftada en az bir kez
temizlenmesi gerekir. Oksijen sağlayıcınız,çalışma koşullarına bağlı olarak filtreleri daha sık
temizlemenizi önerebilir. Hava girişi filtrelerini doğru
şekilde temizlemek için bu adımları uygulayın.
1. İlgili tırnağa bastırıp dışarı çekerek iki filtreyi
FreeStyle Comfort’ın ön kısmından çıkarın.

UYARI: OKSIJEN YOĞUNLAŞTIRICIYI,HAVA GIRIŞI FILTRESI TAKILI OLMADAN
ÇALIŞTIRMAYIN.

PİL TAKIMININ BAKIMI

Pil takımının sıvıya maruz kalmasına izin vermeyin.
Pil takımını sökmeye veya açmaya çalışmayın.
Pili açık aleve maruz bırakmayın.
Pil takımını düşürmeyin.
Pil takımını çocuklardan uzak tutun.
Pil takımınızın zarar gördüğü anlaşılırsa üniteyi
kullanmadan önce oksijen sağlayıcınızla görüşün.
Pili mikrodalga fırın veya başka bir pişirme
cihazının yakınına koymayın.

PİL TAKIMININ SAKLANMASIOksijen yoğunlaştırıcı pilinin şarj edilmedenönce tamamen boşalması gerekmez. Oksijenyoğunlaştırıcı pilinin her kullanımdan sonra şarjedilmesi önerilir.
Pili, doğrudan güneş ışığı veya bir otomobilin
içi gibi aşırı sıcaklığa maruz kalacağı bir yerde
bırakmayın. Bunu yapmak, pilin performansınındüşmesine ve ömrünün kısalmasına yol açabilir.
Ayrıca pilin ısı veya duman yaymasına, tutuşmasına
veya patlamasına yol açabilir. Pil takımı, düşük
nemin (%70'ten düşük bağıl nem) olduğu vepaslandırıcı gazların olmadığı kararlı bir ortamda
saklanmalıdır. Paslanmayı önlemek için ortam
sıcaklığında hızlı dalgalanmalar gibi yoğunlaşmaya
yol açabilecek durumlardan kaçının.
Bir ay veya daha uzun süreli saklama içinboşaltılmış veya %50 ile %75 arası (iki veya
üç LED) kısmen şarj edilmiş durumda saklamaönerilir. Şarj edildikten sonra, pili oksijenyoğunlaştırıcıdan çıkarın.

2. İki filtreyi de ılık sabunlu suda yıkayın.
3. Filtreleri durulayın ve kalan suyu tüy
bırakmayan yumuşak bir havluyla silin. Filtreleri
yerleştirmeden önce filtrelerin tamamen kuruolduğundan emin olun.
4. Temiz ve kuru filtreleri tekrar takın.

PN MN232-C4 L | Kullanıcı El Kitab TR - 427

FreeStyle ComfortTEMİZLEME

Her türlü temizleme işleminden önce üniteyi
KAPALI konuma getirin ve AC veya DC gücübağlantısını kesin. Dış muhafazaya doğrudan ürünPÜSKÜRTMEYİN. Nemli (ıslak olmayan) bir bez
veya sünger kullanın. Kabini ve güç kaynaklarını
temizlemek için beze veya süngere hafif bir ev tipi
temizleyici çözelti püskürtün. Temizlik malzemesiüreticisinin talimatlarına uygun olarak devam edin.
UYARI: ÜNİTEYİ TEMİZLEMEDEN ÖNCE,KAZARA ELEKTRİK ÇARPMASI VE
YANMA TEHLİKESİNİ ÖNLEMEK İÇİNÜNİTEYİ KAPATIN VE GÜÇ KABLOSUNU
ELEKTRİK PRİZİNDEN ÇIKARIN.
YALNIZCA EKİPMAN SAĞLAYICINIZVEYA YETKİLİ BİR SERVİS TEKNİSYENİ
ÜNİTENİN KAPAKLARINI ÇIKARMALIYA DA ÜNİTE ÜZERİNDE SERVİS İŞLEMİ
GERÇEKLEŞTİRMELİDİR.
UYARI: OKSİJEN YOĞUNLAŞTIRICININ VE PİL
TAKIMININ ISLANMAMASINA VE ÜNİTEYE SIVIGİRMEMESİNE ÖZEN GÖSTERİLMELİDİR. BU
DURUM ARIZAYA YA DA KAPANMAYA YOLAÇABİLİR VE ELEKTRİK ÇARPMASI VEYA YANMARİSKİNDE ARTIŞA NEDEN OLABİLİR.
UYARI: DOĞRUDAN ÜNİTE ÜZERİNDE SIVIKULLANMAYIN. İSTENMEYEN KİMYASALMADDELER, BİR SINIRLAMA OLMAKSIZINŞUNLARI İÇERİR: ALKOL VE ALKOL BAZLIÜRÜNLER, KONSANTRE KLOR BAZLI ÜRÜNLER
(ETİLEN KLORÜR) VE YAĞ BAZLI ÜRÜNLER.
ÜNİTEYE ZARAR VEREBİLECEK BU ÜRÜNLER
YOĞUNLAŞTIRICI ÜZERİNDEKİ PLASTİKGÖVDEDE KULLANILMAMALIDIR.
UYARI: KABİN, KONTROL PANELİ VE GÜÇ
KABLOSUNU YALNIZCA NEMLİ (ISLAK OLMAYAN)BİR BEZ VEYA SÜNGER KULLANARAK EV TİPİHAFİF BİR TEMİZLİK MADDESİYLE TEMİZLEYİN
VE ARDINDAN TÜM YÜZEYLERİ KURULAYIN.
YOĞUNLAŞTIRICIYA SIVI GİRMESİNE İZİNVERMEYİN. KANÜL BAĞLANTISININ OKSİJEN
ÇIKIŞININ TOZ, SU VE PARTİKÜLLERDEN ARINMIŞ
OLDUĞUNDAN EMİN OLMAK İÇİN BU ÇIKIŞA
ÖZELLİKLE ÖZEN GÖSTERİN.

428 - TR PN MN232-C4 L | Kullanıcı El Kitab

UYARI: PİL TAKIMINI TEMİZLEMEK İÇİN NEMLİ(ISLAK OLMAYAN) BİR BEZ VEYA SÜNGER
KULLANIN. ÖNCE BEZE VEYA SÜNGERE EV TİPİHAFİF BİR TEMİZLİK MADDESİ PÜSKÜRTÜN VE
PİL TAKIMININ DIŞ MUHAFAZASINI TEMİZLEYİN.
UYARI: OTOMATIK YIKAMA SISTEMIYLETEMIZLEMEYIN.

KANÜLÜN DEĞİŞTİRİLMESİ

Doğru kullanım için daima kanül üreticisinin
talimatlarını izleyin. Burun kanülünü veya
oksijen hortumunu, kanül üreticisinin veya
oksijen sağlayıcınızın önerilerine uygun olaraküreticinin önerisine göre 3 ila 6 ayda bir (her gün
kullanılıyorsa 3 ayda bir) değiştirin. Temizleme
ve değiştirme talimatlarını, hekiminiz veya oksijensağlayıcınız verecektir.
İlave değiştirme kaynakları, oksijen sağlayıcınız
aracılığıyla sağlanır.

FreeStyle Comfort
Alarm DurumlarıÖncelikNo.

Alarm
Açıklaması

1

SistemArızasıYüksek Pil
Sıcaklığı

Ekran

Sesli
SinyalDiğer Ayrıntılar/Önerilen Eylem

Her 20saniyede
1 bip
sesi

Ünite, pil sıcaklığı emniyet kapatma eşiğine
yaklaşırken alarm verecektir.
Kapatma eşiğine ulaşılırsa ünite kapatılacaktır. Harici güç bağlanırsa alarm ekranı
görünecek ve sesli alarm 2 dakika boyunca
devam edecektir.
Pili oda sıcaklığında test edin. Sorun devam ederse hizmet sağlayıcınızla iletişime
geçin.

2

SistemArızasıYüksek
Sıcaklık

Her 20saniyede
1 bip
sesi

Ünite, dahili sıcaklık kapatma eşiğine
yaklaşırken alarm verecektir.
Kapatma eşiğine ulaşılırsa ünite kapatılacak, alarm ekranı görünecek ve sesli alarm
2 dakika boyunca devam edecektir.
Ünitenin deliklerinin herhangi bir şekilde
tıkalı olmadığından emin olun. Sorun
devam ederse hizmet sağlayıcınızla
iletişime geçin.

3

SistemArızasıYüksekBasınç
(Yüksek
SistemBasıncı veya
Kompresör
Arızası)

Her 20saniyede
1 bip
sesi

Kapatma eşiğine ulaşılırsa ünite kapatılacak, alarm ekranı görünecek ve sesli alarm
2 dakika boyunca devam edecektir.

4

SistemArızası AkışHızı (OCD
Valfi Hatası,
Gaz Yolunun
Tıkanması)

Her 30saniyede
1 bip
sesi

Sorun devam ederse hizmet sağlayıcınızla
iletişime geçin.

5Düşük Pil
Nedeniyle
Kapatma

Her 30saniyede
1 bip
sesi

Pil gücü tükenene ve ünite kapanana dek
alarm verilir.

Sorun devam ederse hizmet sağlayıcınızla
iletişime geçin.

Pili şarj edin.

PN MN232-C4 L | Kullanıcı El Kitab TR - 429

FreeStyle ComfortÖncelikNo.

Alarm
Açıklaması

6

Nefes
Algılanmadı

Ekran

Sesli
SinyalDiğer Ayrıntılar

Her 10
dakikada
1 bip
sesi

15 saniye boyunca nefes algılanmadı.
İlk nefesten veya ısınma ekranı kapatıldıktan (hangisi önce gerçekleşirse) sonra
etkinleştirilir.
Kanülde herhangi bir bükülme veya sızıntı
olup olmadığını kontrol edin. Hassasiyet
ayarını artırır. Sorun devam ederse hizmetsağlayıcınızla iletişime geçin.

7

IsınmayokÜnite, nefes algılanana veya ısınma sonaerene (hangisi önce gerçekleşirse) dek
otomatik dozlamayla başlayacaktır.

8

Düşük
Yoğunluk

30
dakikayakadar
her 10
dakikada
1 bip sesi

Sorun devam ederse hizmet sağlayıcınızla
iletişime geçin.

9Düşük Pil
Uyarısı

1 bip
sesi,
tekrarsız

Pili şarj edin.

10

Nefes Hızı
Aşıldı

1 bip
sesi,
tekrarsız

Nefes Hızı 40 bpm’yi aşıyor.

11

SistemArızası Fan
Hatası

yok

Sorun devam ederse hizmet sağlayıcınızla
iletişime geçin.

12

SistemArızası Düşük
Basınç

yok

Sorun devam ederse hizmet sağlayıcınızla
iletişime geçin.

13

SistemArızası Düşük
SıcaklıkyokÜnitenin oda sıcaklığına kadar ısınmasınısağlayın. Sorun devam ederse hizmetsağlayıcınızla iletişime geçin.

430 - TR PN MN232-C4 L | Kullanıcı El Kitab

Doz yanlış bir şekilde tetikleniyorsa hassasiyet ayarını azaltın. Sorun devam ederse
hizmet sağlayıcınızla iletişime geçin.

FreeStyle ComfortÖncelikNo.

Alarm
Açıklaması

BilgiBaşlangıç

Ekran

Sesli
SinyalDiğer Ayrıntılar

1 bilgilendirici bip
sesi

Ekran açılışta görünür ve elektrik zilininçalıştığını onaylayan bir bip sesi duymanız
gerekir.
Başlangıç ekranı, sağ alt köşede geçerli
ürün yazılımı revizyonunu gösterir.

Bilgi

autoSAT
ModundaNormal Çalışma Ekranı(DC Gücü
bağlı değil)

yok

Bilgi

autoSAT
devre dışıyken NormalÇalışmaEkranı (DCGücü bağlı
değil)

yok

Bilgi

Ayar
Değiştirildi

1 bilgilendirici bip
sesi

Bilgi

Kapalı durumunda harici
Güç bağlayın

yok

Bilgi

Çalışmadurumunda
harici güçbağlayın

yok

Bilgi

Kapalı
durumunda
harici gücü
kesin

1 bilgilendirici bipsesi

2 saniye boyunca CAIRE ekranı ve
ardından ekran kapatılmadan önce5 saniye boyunca Pil ve Fiş durumu
görünür

2 saniye boyunca CAIRE ekranı görünür
ve ardından ekran kapatılır.

PN MN232-C4 L | Kullanıcı El Kitab TR - 431

FreeStyle ComfortÖncelikNo.

Alarm
Açıklaması

Ekran

Bilgi

Çalışmadurumundaki
haricigücü kesin(pil varken)

1 bilgilendirici bip
sesi

Bilgi

BeklemeModunda –yukarı
veya aşağı
düğmesinebasıldığında,ekran pilive şarj
durumunu
gösterir

yok

Bilgi

Pil gücünde
Uçak Modu

1 uzunbip sesi

Bilgi

Prize
takılıyken
Uçak Modu

1 uzunbip sesi

432 - TR PN MN232-C4 L | Kullanıcı El Kitab

Sesli
SinyalDiğer Ayrıntılar

Ekran, 5 saniye boyunca görünüp kapatılır.

FreeStyle Comfort
Menü Gezinme Talimatları
Bekleme veya Çalışma modunda Menüye
girebilirsiniz. Çalışma Modundayken bir alarm
durumu oluştuğunda menüden otomatik olarakçıkılır. 20 saniye boyunca herhangi bir düğmeye
basılmazsa Menüden otomatik olarak çıkılır.
Bekleme modunda, üniteyi uyandırmak için Menüdüğmesine iki kez basmanız ve ardından Menüye
girmeniz gerekir.
Dört adet Menü ekranı vardır:

Uçak Modu

1. Durum ekranı: Ürün yazılımı revizyon
numarasını, çalışma süresini, pil boşalma
döngülerini gösterir
2. Uçak modu (AÇIK/KAPALI)
3. Hassasiyet: ayarlanabilir (1-5)
4. autoSAT (AÇIK/KAPALI)
HassasiyetÖrnek Durum Ekranı

AYARLAR MENÜSÜNÜ ÇALIŞTIRMA

autoSAT

Ayarlar menüsüne, Çalışma veya Bekleme
modlarındayken Menü/Seç düğmesinebasılarak girilir.
Menülerde gezinmek içindüğmesine basın.

Durum Sayfası

PN MN232-C4 L | Kullanıcı El Kitab TR - 433

FreeStyle Comfort
DURUM SAYFASIFreestyle Comfort'un Durum Sayfası, Donanım
Yazılımı Revizyonunu, Çalışma Saatlerini ve takılı
pildeki pil döngüsü sayısını gösterir.

Uçak Modunu iptal etmek için İPTAL simgesi
görünürken

düğmesine basın.

Uçak Modu Açık Onaylı

Durum SayfasıÇalışma saatleri tam saat cinsinde gösterilir.
Pil Boşalma sayısı, takılı pilin boşaltıldığı
ve yeniden şarj edildiği döngü sayısıdır.

UÇAK MODUFreeStyle Comfort, Uçak Modu kapalı bir şekilde gelir.

Uçak Modunu kapatmak için KAPALI seçeneğine
geçerken yukarıdaki adımları tekrarlayın.
Ekranda "İleri" simgesi görünürken
düğmesinebasıldığında, menü sayfası sonraki ekranseçeneğine geçer.
Uçak Modu, Çalışma Modunda kullanım içindir.
Ünite pille çalıştırılırken Bekleme modunda olması
halinde, Uçak Modu otomatik olarak devre dışıkalacaktır.
Not: Uçak Modu hakkında daha fazla bilgi almak için
Uçak Modunu Etkinleştirme bölümüne bakın.

HASSASIYET MODU
Uçak Modu Ana Ekranı

Bu modu açmak için DEĞİŞTİR simgesi
görünürken
düğmesine basın. Ekrandakideğer yerine mavi renkli "AÇIK" görünür.

Uçak Modu Açık Seçili

Uçak Modunu açmayı onaylamak için KAYDET
simgesi görünürken
düğmesine basın. Ekrandakideğer yerine siyah renkli "AÇIK" görünür.

434 - TR PN MN232-C4 L | Kullanıcı El Kitab

FreeStyle Comfort, fabrikadan Hassasiyet Modu
2’ye ayarlı bir şekilde gönderilir ve normalkoşullarda herhangi bir ayar gerektirmez.
Hassasiyet Ayarı, 1-5 arasında düzenlenebilir.
Burada 1 en düşük hassasiyet, 5 ise en yüksek
hassasiyettir.

2. Ayarda Hassasiyet Modu Ekranı

FreeStyle Comfort
Hassasiyet Modunu ayarlamak için DEĞİŞTİR
simgesi görünürken
düğmesine basın. Ekrandakideğer yerine bir sonraki mavi sayı görünür.

AUTOSAT MODUFreeStyle Comfort, autoSAT Modu açık birşekilde gelir.

Hassasiyet Modu Ayarı 3 Seçili
autoSAT Modu Ana Ekranı

Hassasiyet Modunda 3. Ayarı onaylamak için
KAYDET simgesi görünürken
düğmesine basın.
Ekrandaki değer yerine siyah renkli "3" görünür.

Bu modu kapatmak içi, DEĞİŞTİR simgesi
görünürken
düğmesine basın. Ekrandaki değer
yerine mavi renkli "KAPALI" görünür.

Hassasiyet Modu değişimini iptal etmek için İPTAL
simgesi görünürken

simgesine basın.

autoSAT Modu Kapalı Seçili

Hassasiyet Modu Ayarı 3 Onaylı

autoSAT Modunun kapalı olduğunu onaylamak için
KAYDET simgesi görünürken
düğmesine basın.
Ekrandaki değer yerine siyah renkli "KAPALI"
görünür.
autoSAT Modunu iptal etmek için İPTAL simgesi
görünürken

düğmesine basın.

Hassasiyet Modu Ayarı 5 (Maks) Onaylı

Ekranda "İleri" simgesi görünürken
düğmesinebasıldığında, menü sayfası sonraki ekranseçeneğine geçer.
autoSAT Modu Kapalı Onaylı

autoSAT Modunu açmak için AÇIK seçeneğine
geçerken yukarıdaki adımları tekrarlayın.
Ekranda "İleri" simgesi görünürken
düğmesinebasıldığında, menü sayfası sonraki ekranseçeneğine geçer.
PN MN232-C4 L | Kullanıcı El Kitab TR - 435

FreeStyle Comfort
Sorun Giderme
Sorun

Olası NedenÇözümFreeStyle Comfort alarm verir
ve kesintili bip sesi çalar.

Alarm Durumları tablosuna
bakın.

Alarm Durumları tablosuna bakın.

Oksijen yoğunlaştırıcı
açılmıyor.

Pil boşalmıştır.

Üniteye AC veya DC çıkışıyla güç verin.

Harici güce bağlanmadı.

Harici bağlantıların güvenli olduğundan emin olun.

Genel arıza.

Pili çıkararak ve AC veya DC harici güç
kaynağını keserek gücü tamamen kapatıp açın.
Ardından pili yeniden takın ve gücü kapatıpaçmayı tamamlayın. İsterseniz harici güç
kaynağını yeniden bağlayabilirsiniz.
Ekipman Sağlayıcınızla iletişime geçin ve gerektiği
şekilde başka bir oksijen kaynağına geçiş yapın.

Tüm diğer sorunlar.

Pili çıkararak ve AC veya DC harici güç
kaynağını keserek gücü tamamen kapatıp açın.
Ardından pili yeniden takın ve gücü kapatıpaçmayı tamamlayın. İsterseniz harici güç
kaynağını yeniden bağlayabilirsiniz.
Mevcut olan başka bir oksijen kaynağına geçiş
yapın ve Ekipman Sağlayıcınızla iletişime geçin.

Not: Bazı alarmlar, ünitenin kapatılmasına yol açacaktır.
Not: Tam bir güç kaybı sırasında alarm sistemi sıfırlanır (önceki alarmlar silinir) ve yalnızca güç tekrar verildikten
sonra durum hala devam ediyorsa bir alarm verilir.

436 - TR PN MN232-C4 L | Kullanıcı El Kitab

FreeStyle Comfort
Seyahat Bilgileri

1. Hiçbir harici güç kaynağı bağlı olmadan
YALNIZCA Pil Gücünü kullanarak POC’yi açın.

FreeStyle Comfort, ticari uçaklarda kullanım için
FAA yönergelerini karşılamaktadır. Her türlü
seyahatten (hava, deniz, demiryolu vs.) önce:• Yoğunlaştırıcının iyi ve çalışır durumda
olduğundan emin olun.
• Hareket etmeden en az 48 saat önce seyahatşirketinizle iletişime geçerek, FreeStyle Comfortünitesiyle seyahat edeceğinizi bildirin.
Seyahatlerde ekstra pil taşımanız önerilir. Havayolu
ile seyahat durumunda, yerde geçirilen zamanı(uçuş öncesi giriş, güvenlik vs.) ve aktarma içinbekleme süresini de göz önünde bulundurarak
yeterli pil ömrü sağlayın. Çoğu havayolu şirketi,öngörülen maksimum yolculuk süresinin en az%150'si (uçuş süreleri, uçuş öncesi ve sonrası
yerde geçirilen süre, aktarmalar ve beklenmedik
gecikmeler) kadar bir süre boyunca cihaza güç
verecek kadar yeterli sayıda tam dolu pil takımı
bulundurmanızı gerekli kılar.
Not: Uçuş sırasında ek pillerin güvenli bir taşıma
çantasında taşınması önerilir.

FAA DÜZENLEMELERİ

Yedek pillerin, orijinal satış ambalajına veya bu
mümkün olmadığında terminallerin yalıtılmasıyla,yani açıkta duran terminallerin üstü bantlanarak
veya pillerin her biri ayrı bir plastik poşete ya
da koruyucu bir keseye koyularak ve yalnızca elbagajında taşınarak kısa devrelere karşı ayrı ayrıkorunması gerekir.

2. Aşağıdaki yöntemlerden birini kullanarak Uçak
Modunu etkinleştirin:a. Uçak Modunu etkinleştirmek için Menü
Gezinme Talimatları bölümündeki talimatlarauyun.

b. Uzun bir bip sesi duyana ve Uçak sembolü
görene kadar Yukarı ve Aşağı düğmelerini
5 saniye boyunca birlikte basılı tutun.

UÇAK MODUNU ETKİNLEŞTİRME
Pil Gücü
POC, hiçbir harici güç kaynağı bağlanmadan
YALNIZCA Pil Gücü kullanılarak normal çalışma
prosedürlerine göre uçakta kullanılabilir.
Harici Güç
Uçakta elektrik prizi varsa, çalışır durumdaysa
ve konumu POC güç kaynağı olarak kullanılması
için uygunsa priz, POC pillerinin yedeği olarak
kullanılabilir. Bu durumda, pilden uçaktaki elektrik
gücüne geçişle ilgili olarak aşağıda önerilen
prosedürün uygulanması gerekir:

3. Uçak sembolü göründüğünde POC’yi prize takın.

Not: Uçaklarla ilgili sınırlamalar nedeniyle, piller Uçak
Modunda şarj olmaz.

PN MN232-C4 L | Kullanıcı El Kitab TR - 437

FreeStyle Comfort
AksesuarlarÜNİTENİN TAŞIMA ÇANTASINAYERLEŞTİRİLMESİ

Uygun performans ve güvenlik için yalnızca
listede belirtilen ve oksijen sağlayıcınız aracılığıyla
CAIRE tarafından tedarik edilen aksesuarları
kullanın. Aşağıda listelenenlerin dışında
aksesuarların kullanılması, yoğunlaştırıcınınperformansını ve/veya güvenliğini olumsuz
etkileyebilir.

1. Filtreleri ve pil boşluklarını hizalayarak üniteyiçantanın dibine yerleştirin.

FreeStyle Comfort Standart AksesuarlarıTaşıma Çantası

MI396-1

Sırt Çantası

MI459-1

Omuz Askısı

MI397-1

AC Güç Kaynağı (Tüm Kablolar
Dahildir)

PW036-1SDC Güç Kablosu

CD041-1

Harici Büyük Partikül Filtresi (Sağ)

FI226-1

Harici Büyük Partikül Filtresi (Sol)FI227-1

2. Çantanın dibindeki fermuarı kapatın.

Mevcut Opsiyonel Aksesuarlar
Ek 8 Hücreli Pil Takımı

BT034-1S

16 Hücreli Pil Takımı

BT035-1S

Pil Takımı Masaüstü Şarj Cihazı

BT036-1

Tüm Parçalar için Aksesuar Çantası

MI372-2

Not: Yukarıda belirtildiği durumlarda, ülkelere özel güç
kabloları için ilave seçenekler mevcut olabilir. Sipariş
için alternatif seçenekler gerektiğinde CAIRE veya
oksijen sağlayıcınızla görüşün.

Taşıma Çantası Talimatları
FreeStyle Comfort ünitenizi taşıma çantasınayerleştirmek ve sabitlemek için aşağıdaki
talimatları yerine getirin.

438 - TR PN MN232-C4 L | Kullanıcı El Kitab

OMUZ ASKISININ TAKILMASI
1. Omuz askısı, doğrudan üniteye veya taşıma
çantasına takılabilir.
2. Askıyı, ünite veya çantadaki halkaya geçirin.

FreeStyle Comfort
Bertaraf ve Yeniden Kullanım
FreeStyle Comfort ve tüm bileşenlerini, doğru
şekilde bertaraf edilmesi için mutlaka ekipmansağlayıcınıza iade edin. Pilin doğru bir biçimde eldençıkarılması için şehrinizdeki yerel idari birimlerle deiletişime geçebilirsiniz. Ünite insanlar arasında tekrar
kullanılabilir ancak yeniden işlenmesi için ekipmansağlayıcısına iade edilmelidir.

WEEE ve RoHS
Bu sembol, ekipmanın sahiplerine
ekipmanı kullanım ömrü sonunda Atık Elektrikli
ve Elektronik Ekipmanlar (WEEE) Direktifine
uygun olarak bir geri dönüşüm tesisine iadeetmelerini hatırlatmaya yöneliktir.

Operatörle Doğrudan Temas
Halindeki Malzemeler• Yoğunlaştırıcı Çantası: PC/ABS• Yoğunlaştırıcı Kontrol Paneli:
Polyester EBG7 veya eşdeğeri
• Oksijen Çıkışı, Ağız: Alüminyum• Ünite Etiketi: Leksan• Kablo Konektörleri: Polikarbonat/Vinil klorür• Güç Kablosu: PVC, Metal• Güç Kaynağı: Leksan 940 (Polikarbonat)• Pil Takımı: PC/ABS•P il Takımı, Güç Kaynağı Etiketleri: Polyester film
• Yoğunlaştırıcı Taşıma Çantası: Cordura, Neopren,
Naylon Astar• Askı: Polipropilen, Asetil, Vinil, Cordura,
NeoprenÜrünlerimiz, Tehlikeli Maddelerin Kısıtlanması(RoHS) direktifine uyar. Eser miktardan fazlakurşun veya başka tehlikeli malzeme içermezler.
DİKKAT: Doğru bir imha işlemi için,imha talimatlarıyla ilgili olarak ekipmansağlayıcınız veya yerel idari birimle
iletişime geçin.
DİKKAT: Moleküler süzgeci her zaman tüm federal,
eyalet ve yerel çevre kontrol düzenlemelerine
uygun olarak bertaraf edin.

PN MN232-C4 L | Kullanıcı El Kitab TR - 439

FreeStyle Comfort
EMC Testi
Yönergeler ve Üretici Beyanı — Elektromanyetik Emisyonlar
FreeStyle Comfort ünitesinin, aşağıda belirtilen elektromanyetik ortamda kullanımı amaçlanmıştır. FreeStyle Comfortmüşterisi veya kullanıcısı, ünitenin bu tür bir ortamda kullanılmasını sağlamalıdır.
Emisyon Testi
RF emisyonları
CISPR 11
RF emisyonları
CISPR 11
Harmonik emisyonlar
IEC 61000-3-2
Voltaj dalgalanmaları/
titreme emisyonları
IEC 61000-3-3

Uyumluluk
Grup 1

Elektromanyetik Ortam — Yönerge
FreeStyle Comfort, RF enerjisini yalnızca dahili fonksiyonu içinkullanır. Bu nedenle, RF emisyonları oldukça düşüktür ve bu
emisyonların, yakındaki elektronik ekipmanla herhangi bir parazite
neden olması beklenmez.

Sınıf B
Sınıf A
Uyumludur

440 - TR PN MN232-C4 L | Kullanıcı El Kitab

FreeStyle Comfort, konutlar ve konutsal amaçlarla kullanılan
binalara tedarik sağlayan kamusal düşük voltajlı güç kaynağı
şebekesine bağlı olan tesisler de dahil olmak üzere, tüm tesislerde
kullanım için uygundur.

FreeStyle Comfort
Yönergeler ve Üretici Beyanı — Elektromanyetik Bağışıklık

FreeStyle Comfort ünitesinin, aşağıda belirtilen elektromanyetik ortamda kullanımı amaçlanmıştır. FreeStyle Comfortmüşterisi veya kullanıcısı, ünitenin bu tür bir ortamda kullanılmasını sağlamalıdır.
Bağışıklık testi
IEC 60601 test düzeyi Uyumluluk düzeyi
Elektromanyetik ortam — yönerge
Elektromanyetik
Zemin ahşap, beton veya seramik karo
±8 kV kontak±8 kV kontak
ortam — yönerge
olmalıdır. Zemin sentetik materyalle kap±15 kV hava tahliyesi±15 kV hava tahliyesi
lanmışsa bağıl nem en az %30 olmalıdır.
IEC 61000-4-2
Elektriksel hızlıgeçiş/patlama
IEC 61000-4-4

Dalgalanma
IEC 61000-4-5

güç besleme hatları için
güç besleme hatları için
±2 kVŞebeke gücü, tipik bir profesyonel sağlık
±2 kVtesisinin ve evde bakım ortamlarının
kalitesinde olmalıdır.
giriş/çıkış hatları için
Yok
±1 kV
hattan hatta ±1 kV
AC hatlarında ±2 kV
ortak modhattan hatta ±1 kV

AC hatlarında ±1 kV
diferansiyelŞebeke gücü, tipik bir profesyonel sağlık
tesisinin ve evde bakım ortamlarının
kalitesinde olmalıdır.

Dış mekan G/Ç hatlarında±2 kV ortak mod0,5 döngü (0°, 45°, 90°, 0,5 döngü (0°, 45°, 90°,135°, 180°, 225°, 270°, 135°, 180°, 225°, 270°,315°) için %0 UT315°) için %0 UT
Güç beslemegiriş hatlarında
gerilim düşmeleri,kısa kesintiler vegerilim değişimleri
IEC 61000-4-11

1 döngü (0°) için %0 UT

1 döngü (0°) için %0 UT

25/30 döngü (0°) için
%70 UT (UT’de %30düşüş)

25/30 döngü (0°) için
%70 UT (UT’de %30düşüş)

250/300 döngü (0°) için
%0 UT

250/300 döngü (0°) için
%0 UTŞebeke gücü, tipik bir profesyonel sağlık
tesisinin ve evde bakım ortamlarının
kalitesinde olmalıdır. FreeStyle Comfort
kullanıcısı için şebeke elektriği kesintileri
sırasında çalışmanın sürdürülmesi
gerekiyorsa FreeStyle Comfort ünitesinekesintisiz güç kaynağından veya pilden
güç verilmesi önerilir.

Güç frekansı manyetik alanları,
profesyonel sağlık tesisi ve evde bakım
ortamlarında bulunan tipik bir konumaözgü düzeylerde olmalıdır.
NOT: UT, test düzeyinin uygulanmasından önce AC şebeke voltajını ifade eder.
Güç frekansı(50/60 Hz) manyetik 3 A/malanı IEC 61000-4-8

3 A/m

NOT: Bazı durumlarda, FreeStyle Comfort önemli bir Elektrostatik Boşalmadan sonra sıfırlanabilir ve normalçalışmaya dönmesi için gücün tamamen kapatılıp yeniden açılması gerekebilir. Daha fazla ayrıntı için Sorun
Giderme bölümüne bakın.

PN MN232-C4 L | Kullanıcı El Kitab TR - 441

FreeStyle Comfort
Yönergeler ve Üretici Beyanı — Elektromanyetik Bağışıklık

FreeStyle Comfort ünitesinin, aşağıda belirtilen elektromanyetik ortamda kullanımı amaçlanmıştır. FreeStyle Comfortmüşterisi veya kullanıcısı, ünitenin bu tür bir ortamda kullanılmasını sağlamalıdır.
Bağışıklık testi IEC 60601 test düzeyi
Uyumluluk düzeyi
Elektromanyetik ortam — yönerge
10 VrmTaşınabilir ve mobil RF iletişim
10 Vrmİletilen RF
ekipmanı, kablolar da dahil olmak üzere
150 kHz - 80 MHz
IEC 61000-4-6
FreeStyle Comfort ünitesinin herhangi
bir parçasına, vericinin frekansı içingeçerli denklem kullanılarak hesaplanmış
önerilen ayırma mesafesinden yakın
10 V/m
10 V/mIşınan RF
olmayacak şekilde kullanılmalıdır.
80 MHz - 2700 MHz
IEC 61000-4-3Önerilen ayrım mesafesi

d = 1.2 P
d = 1.2 P = 80 MHz - 800 MHz
d = 1.2 P = 800 MHz - 2700 MHz
Burada P, verici üreticisine göre vat (W)cinsinden vericinin maksimum çıkış gücü
derecesidir ve d, metre (m) cinsindenönerilen ayırma mesafesidir.
Sabit RF vericilerinin, elektromanyetiksaha etüdüyle belirlenen alan kuvvetleri,aher bir frekans aralığında uyumluluk
düzeyinden düşük olmalıdır.bŞu sembolle işaretlenmiş ekipmanın
yakınında parazit oluşabilir:

NOT 1: 80 MHz ve 800 MHz’de, daha yüksek frekans aralığı geçerlidir.
NOT 2: Bu yönergeler her durumda geçerli olmayabilir. Elektromanyetik yayılım; yapılar, nesneler ve insanlar
tarafından gerçekleştirilen absorpsiyondan ve yansımadan etkilenir.
aRadyo frekansı ile çalışan telefonların (cep telefonları/kablosuz telefonlar) baz istasyonları ve hareketli kara
telsizleri, amatör telsizler, AM ve FM radyo yayını ve TV yayını gibi sabit vericilerden kaynaklanan alan kuvvetleri,teorik olarak doğru bir biçimde tahmin edilemez. Sabit RF vericilerinden kaynaklanan elektromanyetik ortamıdeğerlendirmek için bir elektromanyetik saha araştırmasının yapılması göz önünde bulundurulmalıdır. FreeStyle
Comfort ünitesinin kullanıldığı konumdaki alan kuvveti, yukarıdaki geçerli RF uyumluluk düzeyini aşıyorsa FreeStyle
Comfort normal çalıştığının doğrulanması için gözlemlenmelidir. Anormal performans gözlemlenirse, FreeStyle
Comfort ünitesinin yönünün veya konumunun değiştirilmesi gibi ek önlemler gerekli olabilir.
b

150 kHz ile 80 MHz frekans aralığında alan kuvvetleri 10 V/m’den düşük olmalıdır.

442 - TR PN MN232-C4 L | Kullanıcı El Kitab

FreeStyle ComfortTaşınabilir ve hareketli RF haberleşme ekipmanları ile FreeStyle Comfort arasında
bırakılması önerilen mesafeler

FreeStyle Comfort ünitesinin, ışınan RF bozulmalarının kontrol altında olduğu bir elektromanyetik alanda kullanımı
amaçlanmıştır. FreeStyle Comfort müşterisi veya kullanıcısı, elektromanyetik paraziti önlemeye yardımcı olmak
için taşınabilir ve hareketli RF haberleşme ekipmanları (vericiler) ile FreeStyle Comfort arasında, haberleşme
ekipmanının maksimum çıkış gücüne uygun olarak aşağıda belirtildiği gibi bir minimum mesafe bırakabilir.
Vericinin nominal
Vericinin frekansına göre ayrım mesafesi
maksimum çıkış gücüm
150 kHz - 80 MHz

80 MHz - 800 MHz

800 MHz - 2,5 GHzW

d = 1.2 P
d = 1.2 P
d = 1.2 P
0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
Yukarıda listelenmeyen bir nominal maksimum çıkış gücüne sahip olan vericiler için metre (m) cinsinden önerilen
ayrım mesafesi d, vericinin frekansı için geçerli olan denklem kullanılarak tahmin edilebilir; burada P, vericiüreticisine göre vat (W) cinsinden vericinin maksimum çıkış gücü derecelendirmesidir.
NOT 1: 80 MHz ve 800 MHz’de, daha yüksek frekans aralığına yönelik ayrım mesafesi geçerlidir.
NOT 2: Bu yönergeler her durumda geçerli olmayabilir. Elektromanyetik yayılım; yapılar, nesneler ve insanlar
tarafından gerçekleştirilen absorpsiyondan ve yansımadan etkilenir.

Sınıflandırma

Elektrik çarpmasına karşı koruma türü:
Sınıf II	Elektrik çarpmasına karşı koruma			çift izolasyonla elde edilir.
Elektrik çarpmasına karşı koruma derecesi:Şunlar bakımından elektrik çarpmasına karşı belirli
bir derecede koruma sağlayan BF Tipi Ekipman:		1) izin verilen kaçak akım;
		2) koruyucu topraklama bağlantısının
güvenilirliği (varsa).
		Doğrudan kardiyak uygulama içintasarlanmamıştır.
Tıbbi Elektrikli Ekipman Standardı için bağımsız test:	Şu standartlara uygunluğu test edilmiştir:• IEC 60601-1 sürüm 3.1: Tıbbi Elektrikli
ekipman — Bölüm 1 Genel Güvenlik
Gereksinimleri.
• CAN/CSA C22.2 No. 60601-1-8 sürüm 2.1,
Tıbbi Elektrikli Ekipman — Bölüm 1:
Genel Güvenlik Gereksinimleri ile uyumlu
bulunmuştur
Ekipman ile diğer cihazlar arasında potansiyel
elektromanyetik veya diğer parazite karşı koruma.
• EN 60601-1-2/IEC 60601-1-2: 4. Sürüm ile
uyumlu olarak test edilmiştir.

	• RTCA/DO160 ile uyumlu olarak test edilmiştir
	Bölüm 21 Kategori M.
• CISPR 11/EN 55011 Sınıf B Grup 1,
	"Endüstriyel, Bilimsel ve Tıbbi (ISM)	Ekipman"• FCC Bölüm 15, Alt Bölüm B — Sınıf B	İstenmeden Işın Yayanlar
IP22 — Parmakların veya çapı 12,5 mm’den büyük
olan nesnelerin girişine karşı koruma. Dikey olarak damlayan su, muhafaza normal konumundan
15°’lik bir açıyla eğildiğinde zararlı bir etkiye
neden olmaz.
İzin verilen temizlik ve enfeksiyon kontrolü
yöntemi: Lütfen bu FreeStyle Kullanıcı
El Kitabının "Temizlik, Bakım ve Uygun
Servis İşlemleri" bölümünü inceleyin.
Yanıcı anestezik gazların mevcut olduğu ortamda
uygulama güvenliği derecesi:	Ekipman bu tür bir uygulama için uygun değildir.
Çalışma modu: Sürekli çalışma.

PN MN232-C4 L | Kullanıcı El Kitab TR - 443

FreeStyle ComfortUživatelské ovládací prvky a ukazatele stavu systému
ISO 7000Příručka pro obsluhu; návod
k obsluze. Reg. č. 1641
Země a datum výroby. „CC“ označuje
dvoupísmenný kód země výroby.
Datum výroby je ve formátu
RRRR-MM-DD. Reg. č. 6049

ISO 7010: Grafické značky – bezpečnostníbarvy a bezpečnostní značky – registrovanébezpečnostní značky
Je nezbytné přečíst příručku
k obsluze. Reg. č. M002Chraňte před otevřeným plamenem,ohněm, jiskrami. Zákaz kouření
a manipulace s plamenem.
Reg. č. P003

Chraňte před deštěm, uchovávejte
v suchu. Reg. č. 0626

V blízkosti jednotky ani při
manipulaci s ní nekuřte. Reg. č. P002

Limit stohování podle počtu, kde „n“označuje maximální povolený počet
jednotek. Reg. č. 2403

Varování. Reg. č. W001

Katalogové číslo. Reg. č. 2493
Sériové číslo. Reg. č. 2498

IEC 60417
Příložná část typu BF (stupeň ochrany
proti úrazu elektrickým proudem).
Reg. č. 5333

Rozsah skladovací nebo provozní
teploty. Reg. č. 0632
Rozsah skladovací vlhkosti.
Reg. č. 2620
Omezení atmosférického tlaku.
Reg. č. 2621
Název a adresa výrobce. Reg. č. 3082
Touto stranou nahoru. Reg. č. 0623
Křehké, manipulujte opatrně.
Reg. č. 0621

Zařízení třídy II s dvojitou izolací.
Reg. č. 5172
Nařízení EU 2017/745
Zplnomocněný zástupce v Evropském
společenství
Pokud je na štítku jedinečného
identifikátoru prostředku (UDI)
uveden symbol CE####, prostředek
splňuje požadavky týkající sezdravotnických prostředků. Symbol
CE#### označuje číslo oznámeného
subjektu.
Další symbolyChraňte před hořlavými materiály,
oleji a mazivy.

Obsahuje nebezpečné látky.
Reg. č. 3723

Nerozebírejte.

Dovozce. Reg. č. 3725

Zvyšte průtok.

Výstup O2

Snižte průtok.
Tlačítko Menu.
Vypínač. Reg. č. 5009
V souladu s
TCA/DO-160,
oddíl 21,kategorie M

444- CZ PN MN232-C4 L | Uživatelská příručkaVýrobce tohoto POC stanovil, že totozařízení splňuje všechna příslušná
kritéria přijatelnosti FAA a použití
na palubě letadla.

FreeStyle ComfortVýrobce tohoto POC stanovil, že totozařízení splňuje všechna příslušná
kritéria přijatelnosti FAA a použití napalubě letadla.
Symbol recyklace
Autorizovaný zástupce ve Švýcarsku(PŘIDAT)
Pokud je zařízení označeno značkou
UKCA, jak je znázorněno s číslem
UKCA#### označujícím číslo
oznámeného subjektu, je toto zařízení
v souladu s předpisy UKCA.
21 CFR 801.15
Tato zařízení smí být podle
federálních zákonů prodávána pouzelékaři nebo na jejich příkaz.
Certifikace ETL
Certifikováno pro americký
i kanadský trh podle platných
amerických a kanadských norem.
Obsahuje FCC ID: WAP2001
Vyhláška FCC
Tento přístroj může obsahovat modul
CYBLE-022001-00, včetně antény 2450AT18B100
od společnosti Johnson Technology, splňuje
část 15 předpisů FCC. Zařízení splňuje
požadavky na modulární vysílače schválené
podle veřejného oznámení FCC DA00-1407.
Provoz vysílače podléhá následujícím dvěma
podmínkám: (1) Zařízení nesmí způsobovatškodlivé rušení a (2) Toto zařízení musí akceptovat
veškeré přijaté rušení, včetně rušení, které můžezpůsobit nežádoucí provoz. Na výrobek se můževztahovat jeden nebo více patentů, amerických
i mezinárodních. Seznam platných patentů
najdete na níže uvedené webové stránce.
Pat.: www.caireinc.com/corporate/patents
Grafické symboly pro použití na zařízeních
IEC 60417
Stejnosměrný proud Reg. č. 5031

IEC 60601-1
Chráněno proti kapající voděSměrnice Rady 2012/19/EU
OEEZ
Tento symbol má majitelům zařízení
připomenout, že jej mají podlesměrnice o odpadních elektrických
a elektronických zařízeních (OEEZ)
na konci životnosti odevzdat
sběrnému recyklačnímu zařízení.
Naše výrobky odpovídají směrnici
o omezení používání některých
nebezpečných látek v elektrických
a elektronických zařízeních (RoHS).
Neobsahují více než stopové množství
olova nebo jiných nebezpečných
materiálů.
IEC 60601-1-8
Technický alarm nízké priority
ISO 15223-1
Zdravotnický prostředek.
Reg. č. 5.7.7
Jedinečný identifikátorzařízení č. 5.7.10
A S/NZS 4417.1, AS/NZS 4417.2
Značka ACMA Regulatory
Compliance – označuje soulads předpisy ACMA (ADD)
Bezdrátová technologie
Typy spojení

SSP, iAP2, GATTČetnost2 402 až 2 480 MHz

Max. RF výkon

+4 dBm

Provozní rozsah

10 m (třída 2)

Doporučujeme, aby během provozu byl přístroj vevzdálenosti minimálně 1,1 cm (0,43 in.) od těla.
Nevztahuje se na masky, vzduchovou hadičku nebopříslušenství.

Střídavý proud Reg. č. 5032

PN MN232-C4 L | Uživatelská příručka CZ - 445

FreeStyle Comfort
Rozbalení vašeho přístroje FreeStyle Comfort
Zkontrolujte, zda jsou v baleníobsaženy všechny uvedené
součásti. Pokud některépoložky chybí, okamžitě
kontaktujte dodavatele kyslíku.

Standardní příslušenství
Volitelné příslušenství• Přenosný kyslíkový koncentrátor • Záložní 8článková baterie• Přenosná taška• 16článková baterie• Ramenní popruh• Stolní nabíječka• 8článková baterie• Přenosná taška pro veškeré
příslušenství• Střídavý napájecí zdroj•N apájecí kabel pro střídavý proud• Napájecí zdroj pro stejnosměrný
proud• Uživatelská příručka

Seznámení s přístrojem FreeStyle Comfort
Seznamte se s hlavními prvky přístroje FreeStyle Comfort a jeho uživatelským rozhraním.

Čelní pohled
Poutko na popruhUživatelské
rozhraníVýstupkyslíkuVýstupvzduchuPřívodvzduchu /filtryPřívod vzduchu /filtryKryt samice
napájecího
konektoru
BaterieTlačítko pro
vysunutí baterie
446- CZ PN MN232-C4 L | Uživatelská příručka

FreeStyle Comfort
Horní pohled
Obrazovka
displejeZvýšení průtoku

Tlačítko vypínače

Tlačítko MenuSnížení průtoku

Obrazovka displeje při použití baterie
Nastavení průtoku

Zobrazení baterie

Ukazatel autoSAT
ZobrazenívýstrahUkazatelpodání boluZbývajícíprocento nabitíbaterieObrazovka displeje při použití napájecího zdroje pro střídavý i stejnosměrný proud

Nastaveníprůtoku

Prostor pro
zobrazenívýstrah

Zobrazení baterie

Ukazatel dobíjení

PN MN232-C4 L | Uživatelská příručka CZ - 447

FreeStyle Comfort
Co je kyslíkový koncentrátor
Vzduch, který dýcháme, obsahuje přibližně
21 % kyslíku, 78 % dusíku a 1 % dalších plynů.
U FreeStyle® Comfort® je vzduch z místnosti
nasáván do přístroje přes přívody vzduchu.
Následně prochází přes adsorpční materiál zvaný
molekulové síto. Tento materiál oddělí kyslík
od dusíku a umožní pouze průchod kyslíku.
Výsledkem je proud vysoce koncentrovaného
kyslíku dodávaný uživateli.
Přístroj FreeStyle Comfort kombinuje vyspělý
kyslíkový koncentrátor s technologií hospodaření
s kyslíkem za účelem vytvoření přenosnéhozařízení, které umožní maximální přenosnostpři chůzi. Pokrokový systém dodávání kyslíku
prostřednictvím pulzního proudění rychle rozezná,
kdy se uživatel nadechuje, a na počátku každého
nádechu poskytne proud kyslíku. Dodávka pulzním
prouděním je ideální pro různé aktivity a dobu
pobytu mimo domov. Na rozdíl od kontinuálníhoprůtoku lépe hospodaří s kyslíkem a prodlužuje
životnost baterie. Přístroj FreeStyle Comfort je taklehký, že je možné jej přenášet pomocí popruhu
na rameno. Může být napájen střídavým proudem(ze zásuvky) i stejnosměrným proudem (ve vozidle)
nebo bateriemi.

Určený účel použití
Kyslíkový koncentrátor CAIRE FreeStyle Comfort
je určen k podávání doplňkového kyslíku. Přístroj
není určen k podpoře života ani neposkytuje žádné
možnosti sledování pacienta.

Proč vám lékař předepsal kyslík
Mnoho lidí trpí různými srdečními, plicními
a jinými dýchacími onemocněními. Významnémnožství těchto lidí může profitovat z dodatečné
léčby kyslíkem doma, na cestách nebo přikaždodenních aktivitách mimo domov.
Vzduch, který dýcháme, je z 21 % tvořen právě
tímto plynem, kyslíkem. Aby lidské tělo správně
fungovalo, potřebuje jeho nepřetržitý přísun.
Lékař vám předepsal míru proudění nebo nastavenípřístroje podle konkrétního onemocnění dýchacíhoústrojí.

448- CZ PN MN232-C4 L | Uživatelská příručkaPřestože kyslík není návyková látka, neoprávněná
léčba kyslíkem může být nebezpečná.
Před použitím tohoto kyslíkového koncentrátoru
se musíte poradit s lékařem. Dodavatel kyslíkovéhopřístroje vám názorně ukáže způsob nastavenípředepsaného průtočného množství.

Profil uživatele
Koncentrátory jsou určeny k dodávce doplňkového
kyslíku uživatelům, kteří trpí obtížemi v důsledku
onemocnění ovlivňujícího účinnost jedné z plicpři přenosu kyslíku ze vzduchu do krevníhooběhu. Přenosné kyslíkové koncentrátory (PKK)
neskladují ani neobsahují kyslík. Není potřeba
je doplňovat a lze je dobít na každém místě
s dostupným střídavým nebo stejnosměrným
napájením (kromě letadel). Díky tomu je uživatel
relativně soběstačný z hlediska domácího použití,přecházení (uvnitř domu i venku), mobility
a celkového životního stylu. Použití kyslíkového
koncentrátoru vyžaduje lékařský předpis.
Typický uživatel léčby kyslíkem je starší 65 let
a trpí různými respiračními chorobami, včetně
chronické obstrukční plicní nemoci (CHOPN).
Uživatelé mají obvykle dobré kognitivní schopnosti
a musí být schopni informovat o bolestech. Pokuduživatel není schopen informovat o bolestech
nebo nedokáže přečíst a chápat značení a pokyny
k použití koncentrátoru, doporučuje se použití
koncentrátoru pouze pod dohledem osoby, která
je toho schopná. Pokud uživatelé při použití
koncentrátoru cítí nějaké bolesti, doporučuje se,
aby kontaktovali svého poskytovatele zdravotnípéče. Zároveň se uživatelům doporučuje, abyměli k dispozici záložní kyslík (tj. kyslíkovou
láhev) pro případ výpadku proudu nebo poruchy
koncentrátoru. K použití koncentrátoru nejsou
nutné žádné další jedinečné dovednosti aniuživatelské schopnosti.

FreeStyle ComfortDůležité!
Bezpečnostní pokyny jsou určeny následujícímzpůsobem:
VAROVÁNÍ: DŮLEŽITÉ BEZPEČNOSTNÍ
INFORMACE O NEBEZPEČÍCH, KTERÁMOHOU ZPŮSOBIT VÁŽNÁ ZRANĚNÍ.
UPOZORNĚNÍ: Důležité informace,
jak zabránit poškození přístroje
FreeStyle Comfort.
Poznámka: Informace, které vyžadují zvláštní pozornost.

Indikace pro použití
VAROVÁNÍ: JE VELMI DŮLEŽITÉ ZVOLITPOUZE PŘEDEPSANOU ÚROVEŇ
KYSLÍKU. NEMĚŇTE VÝBĚR PRŮTOKU,POKUD VÁM TO NEBYLO NAŘÍZENO
OŠETŘUJÍCÍM LÉKAŘEM.
VAROVÁNÍ: TOTO ZAŘÍZENÍ NEBYLO TESTOVÁNOANI SCHVÁLENO PRO POUŽITÍ U PEDIATRICKÝCH
PACIENTŮ.
VAROVÁNÍ: V USA SMÍ BÝT PODLE FEDERÁLNÍCH
ZÁKONŮ TATO ZAŘÍZENÍ PRODÁVÁNA
A PRONAJÍMÁNA POUZE LÉKAŘI ČI JINÝMI
LICENCOVANÝMI POSKYTOVATELI ZDRAVOTNÍ
PÉČE NEBO NA JEJICH PŘÍKAZ.
VAROVÁNÍ: TATO JEDNOTKA NENÍ URČENA
K POUŽITÍ PRO PODPORU ŽIVOTA. U KAŽDÉHO
UŽIVATELE, KTERÝ NENÍ SCHOPEN SDĚLIT
NEPOHODLÍ PŘI POUŽÍVÁNÍ TOHOTO ZAŘÍZENÍ,MŮŽE BÝT VYŽADOVÁNO DALŠÍ SLEDOVÁNÍ.
UŽIVATELÉ S POŠKOZENÝM SLUCHEM A/NEBOZRAKEM MOHOU POTŘEBOVAT POMOC PŘI
SLEDOVÁNÍ ALARMŮ.
VAROVÁNÍ: NASTAVENÍ REŽIMU PULZNÍCH
DÁVEK MUSÍ BÝT STANOVENO PRO KAŽDÉHO
UŽIVATELE INDIVIDUÁLNĚ PODLE JEHOVLASTNÍCH POTŘEB BĚHEM KLIDU, CVIČENÍ
A CESTOVÁNÍ.
VAROVÁNÍ: TOTO ZAŘÍZENÍ JE URČENO
K POUŽITÍ V DOMÁCÍM ZDRAVOTNICKÉMA PODOBNÉM AMBULANTNÍM PROSTŘEDÍ.

VAROVÁNÍ: NĚKTERÉ POKUSY PACIENTAO DÝCHÁNÍ NEMUSÍ SPUSTIT HOSPODÁRNOUFUNKCI.
VAROVÁNÍ: PŘÍSTROJ FREESTYLE COMFORTNENÍ URČEN K POUŽITÍ U PACIENTŮ, KTEŘÍ
PRODĚLALI TRACHEOTOMII.
VAROVÁNÍ: TĚHOTNÉ NEBO KOJÍCÍ ŽENYBY NEMĚLY POUŽÍVAT PŘÍSLUŠENSTVÍDOPORUČENÉ V TÉTO PŘÍRUČCE,JELIKOŽ MŮŽE OBSAHOVAT FTALÁTY.

Kontraindikace pro použití
VAROVÁNÍ: ZA URČITÝCH OKOLNOSTÍ
MŮŽE BÝT POUŽITÍ NEPŘEDEPSANÉHOKYSLÍKU NEBEZPEČNÉ. TOTOZAŘÍZENÍ BY MĚLO BÝT POUŽÍVÁNO
POUZE TEHDY, POKUD BYLO
PŘEDEPSÁNO LÉKAŘEM.
VAROVÁNÍ: TOTO ZAŘÍZENÍ NENÍ URČENO
PRO POUŽITÍ V PROSTŘEDÍ NEBO V BLÍZKOSTIPROSTŘEDÍ MAGNETICKÉ REZONANCE, JAKO
JSOU PŘÍSTROJE MR.
VAROVÁNÍ: NENÍ URČEN K POUŽITÍ
V PŘÍTOMNOSTI HOŘLAVÝCH ANESTETIK.
VAROVÁNÍ: STEJNĚ JAKO U JINÝCH ELEKTRICKYNAPÁJENÝCH ZAŘÍZENÍ MŮŽE DOJÍT K SITUACI,KDY PŘÍSTROJ NEBUDE MOCI FUNGOVATKVŮLI PŘERUŠENÍ NAPÁJENÍ, PŘÍPADNĚ
BUDE NUTNÉ PROVÉST SERVIS KYSLÍKOVÉHOKONCENTRÁTORU KVALIFIKOVANÝMTECHNIKEM. KYSLÍKOVÝ KONCENTRÁTOR
NENÍ VHODNÝ PRO ŽÁDNÉHO UŽIVATELE,JENŽ BY BYL KVŮLI TOMUTO DOČASNÉMUPŘERUŠENÍ VYSTAVEN NEPŘÍZNIVÝM
DŮSLEDKŮM NA ZDRAVÍ.
VAROVÁNÍ: TOTO ZAŘÍZENÍ NENÍ SCHVÁLENO
PRO POUŽITÍ VE VŠECH PROSTŘEDÍCH.
VYSTAVENÍ URČITÝM ELEKTROMAGNETICKÝMZDROJŮM MŮŽE VÉST K RUŠENÍ NEBO
NEZAMÝŠLENÉMU VÝKONU ZAŘÍZENÍ.
V PŘÍPADĚ RUŠENÍ JE TŘEBA ZAŘÍZENÍ
OKAMŽITĚ VYPNOUT.

VAROVÁNÍ: TOTO ZAŘÍZENÍ JE URČENO
K POUŽITÍ PACIENTEM, KTERÉMUJE PODÁVÁNA KYSLÍKOVÁ TERAPIE.

PN MN232-C4 L | Uživatelská příručka CZ - 449

FreeStyle Comfort
Bezpečnostní pokyny
VAROVÁNÍ: NEJSOU POVOLENYŽÁDNÉ ÚPRAVY TOHOTO ZAŘÍZENÍ.
VAROVÁNÍ: TENTO PŘÍSTROJ DODÁVÁ VYSOCE
KONCENTROVANÝ KYSLÍK, KTERÝ ZNAČNĚ
PODPORUJE HOŘENÍ. VYVARUJTE SE KOUŘENÍ
NEBO OTEVŘENÝCH PLAMENŮ VE STEJNÉMÍSTNOSTI, KDE SE NACHÁZÍ (1) TENTO
PŘÍSTROJ NEBO (2) JAKÉKOLI PŘÍSLUŠENSTVÍ
PŘENÁŠEJÍCÍ KYSLÍK. NEDODRŽENÍ TOHOTOVAROVÁNÍ MŮŽE VÉST K VÁŽNÉMU POŽÁRUA ŠKODÁM NA MAJETKU A/NEBO ZPŮSOBITZRANĚNÍ ČI SMRT.
VAROVÁNÍ: NEPOUŽÍVEJTE JEDNOTKUV OMEZENÉM NEBO STÍSNĚNÉM PROSTORU(TJ. V MALÉM POUZDRU NEBO RUČNÍ TAŠCE),V NĚMŽ MŮŽE BÝT OMEZENO VĚTRÁNÍ.
MOHLO BY DOJÍT K PŘEHŘÁTÍ KYSLÍKOVÉHOKONCENTRÁTORU A SNÍŽENÍ JEHO VÝKONU.
VAROVÁNÍ: KONCENTRÁTOR BY MĚL BÝTUMÍSTĚN TAK, ABY BYL CHRÁNĚN PŘEDKOUŘEM, ZNEČIŠTĚNÍM ČI VÝPARY.
VAROVÁNÍ: V PŘÍPADĚ, ŽE DOJDE K ZÁVAŽNÉ
UDÁLOST S TÍMTO PŘÍSTROJEM, MĚL BY
UŽIVATEL TUTO UDÁLOST NEPRODLENĚ
NAHLÁSIT POSKYTOVATELI A/NEBO VÝROBCI.
ZÁVAŽNÁ UDÁLOST JE DEFINOVÁNA JAKO
ZRANĚNÍ, SMRT NEBO MOŽNOST ZPŮSOBITZRANĚNÍ/SMRT V PŘÍPADĚ OPAKOVANÉHOVÝSKYTU UDÁLOSTI. UŽIVATEL MŮŽE TAKÉNAHLÁSIT UDÁLOST PŘÍSLUŠNÉMU ORGÁNUV ZEMI, KDE K UDÁLOSTI DOŠLO.
VAROVÁNÍ: POUŽITÍ URČITÝCH
PŘÍSLUŠENSTVÍ PRO VEDENÍ KYSLÍKU,KTERÁ NEJSOU URČENA PRO TENTO
KYSLÍKOVÝ KONCENTRÁTOR, MŮŽEVÉST KE SNÍŽENÍ JEHO VÝKONU.
DOPORUČENÁ PŘÍSLUŠENSTVÍ JSOUUVEDENA V TÉTO PŘÍRUČCE.
VAROVÁNÍ: POKUD BYL KYSLÍKOVÝ KONCENTRÁTOR UPUŠTĚN, POŠKOZEN NEBO VYSTAVENPŮSOBENÍ VODY, OBRAŤTE SE NA DODAVATELE
ZAŘÍZENÍ, ABY PŘÍSTROJ ZKONTROLOVAL NEBO
PŘÍPADNĚ OPRAVIL. NEPOUŽÍVEJTE KYSLÍKOVÝ
KONCENTRÁTOR, POKUD MÁ POŠKOZENÝ NAPÁJECÍ KABEL NEBO ZÁSTRČKU.
VAROVÁNÍ: PŘI PŘEPRAVĚ VE VOZIDLE KYSLÍKOVÝ
KONCENTRÁTOR PEČLIVĚ PŘIPOUTEJTE NEBO
JINÝM ZPŮSOBEM ZAJISTĚTE, ABYSTE ZABRÁNILI
POŠKOZENÍ NEBO ZRANĚNÍ.
450- CZ PN MN232-C4 L | Uživatelská příručka

VAROVÁNÍ: PŘEDPISY MINISTERSTVA DOPRAVY
USA (DOT) A ORGANIZACE SPOJENÝCH NÁRODŮ
(OSN) VYŽADUJÍ PŘI VŠECH MEZINÁRODNÍCHLETECH VYJMUTÍ BATERIE ZE ZAŘÍZENÍ
V PŘÍPADĚ, ŽE JE KYSLÍKOVÝ KONCENTRÁTORODBAVEN JAKO ZAVAZADLO. TAKÉ PŘIPŘEPRAVĚ KYSLÍKOVÉHO KONCENTRÁTORU
MUSÍ BÝT BATERIE ZE ZAŘÍZENÍ VYJMUTAA ŘÁDNĚ ZABALENA.
VAROVÁNÍ: TENTO PŘÍSTROJ NEPOUŽÍVEJTE,DOKUD SI NEPŘEČTETE A NEPOCHOPÍTE TUTOPŘÍRUČKU. POKUD NEROZUMÍTE VAROVÁNÍMA POKYNŮM, KONTAKTUJTE DODAVATELE
ZAŘÍZENÍ, DŘÍVE NEŽ PŘÍSTROJ POUŽIJETE,JINAK MŮŽE DOJÍT KE ZRANĚNÍ NEBO ŠKODÁM.
VAROVÁNÍ: POKUD CÍTÍTE BOLESTI NEBO SE
NACHÁZÍTE V NALÉHAVÉM ZDRAVOTNÍM STAVU,OKAMŽITĚ VYHLEDEJTE LÉKAŘSKOU POMOC.
VAROVÁNÍ: PROVOZ KYSLÍKOVÉHOKONCENTRÁTORU MIMO URČENÝ ROZSAHPROVOZNÍCH TEPLOT MŮŽE OMEZITSCHOPNOST KONCENTRÁTORU SPLNIT
SPECIFIKACE KYSLÍKOVÉHO KONCENTRÁTORU.
MEZNÍ HODNOTY TEPLOTY NAJDETE V ČÁSTI
TÉTO PŘÍRUČKY VĚNOVANÉ SPECIFIKACÍM.
VAROVÁNÍ: NA PŘÍSLUŠENSTVÍ PŘENÁŠEJÍCÍ
KYSLÍK A NA KYSLÍKOVÝ KONCENTRÁTOR
NEPOUŽÍVEJTE OLEJE, MAZIVA, MATERIÁLY
NA BÁZI ROPY ANI JINÉ HOŘLAVÉ VÝROBKY.
POUŽÍVAT BY SE MĚLY POUZE ROZTOKYA BALZÁMY NA BÁZI VODY, KTERÉ JSOU
KOMPATIBILNÍ S KYSLÍKEM. KYSLÍK URYCHLUJESPALOVÁNÍ HOŘLAVÝCH LÁTEK.
VAROVÁNÍ: NESPRÁVNÉ POUŽITÍ BATERIEMŮŽE ZPŮSOBIT, ŽE SE BATERIE ZAHŘEJEA VZNÍTÍ, COŽ MŮŽE VÉST K VÁŽNÝM
ZRANĚNÍM. DEJTE POZOR, ABYSTE BATERIINEPROPÍCHLI, NEUDEŘILI S NÍ, NESTOUPLI NA NIČI JI NEUPUSTILI A ABYSTE JI NEVYSTAVOVALIJINÝM SILNÝM ÚDERŮM NEBO NÁRAZŮM.
POUŽITÍ POŠKOZENÉ BATERIE MŮŽE ZPŮSOBITZRANĚNÍ OSOB.
VAROVÁNÍ: NEZKRATUJTE KOVOVÉ KONTAKTY
BATERIE KOVOVÝMI PŘEDMĚTY, JAKO JSOU
KLÍČE NEBO MINCE. TO MŮŽE ZPŮSOBIT JISKRYNEBO VÉST K TVORBĚ NADMĚRNÉHO TEPLA.

FreeStyle Comfort
VAROVÁNÍ: PŘI POUŽITÍ PŘENOSNÉHOKYSLÍKOVÉHO KONCENTRÁTORU VE
VENKOVNÍM PROSTŘEDÍ S NAPÁJENÍM
STŘÍDAVÝM PROUDEM ZAPOJTE NAPÁJENÍ
VŽDY POUZE DO ZÁSUVKY S OCHRANOU PROTIZEMNÍMU SPOJENÍ (GFI).
VAROVÁNÍ: KYSLÍKOVÝ KONCENTRÁTOR BY
SE NEMĚL POUŽÍVAT V BLÍZKOSTI JINÝCH
ZAŘÍZENÍ ANI BÝT ULOŽEN NA NICH ČI POD NIMI.
POKUD SE TAKOVÉMU POUŽITÍ NELZE VYHNOUT,ZAŘÍZENÍ BY MĚLO BÝT SLEDOVÁNO, ABY SEOVĚŘIL JEHO BĚŽNÝ PROVOZ.
VAROVÁNÍ: POUŽITÍ JINÝCH NEŽ K TOMUURČENÝCH KABELŮ A ADAPTÉRŮ, S VÝJIMKOUKABELŮ A ADAPTÉRŮ PRODÁVANÝCH
VÝROBCEM ZDRAVOTNICKÉHO ELEKTRICKÉHOPŘÍSTROJE JAKO NÁHRADNÍ DÍLY ZA VNITŘNÍ
SOUČÁSTI, MŮŽE VÉST KE SNÍŽENÍ ODOLNOSTIKYSLÍKOVÉHO KONCENTRÁTORU.
VAROVÁNÍ: POUŽÍVEJTE POUZE ELEKTRICKÉNAPĚTÍ UVEDENÉ NA SPECIFIKAČNÍM ŠTÍTKU
PŘIPEVNĚNÉM NA PŘÍSTROJI.
VAROVÁNÍ: V KOMBINACI S TÍMTO VÝROBKEM
NEPOUŽÍVEJTE PRODLUŽOVACÍ KABELY ANINEZAPOJUJTE PŘÍLIŠ MNOHO ZÁSTRČEK
DO STEJNÉ ELEKTRICKÉ ZÁSUVKY. POUŽITÍPRODLUŽOVACÍCH KABELŮ NEPŘÍZNIVĚ
OVLIVŇUJE VÝKON ZAŘÍZENÍ. PŘÍLIŠ MNOHOZÁSTRČEK V JEDNÉ ZÁSUVCE MŮŽE VÉSTK PŘETÍŽENÍ ELEKTRICKÉHO PANELU, COŽZPŮSOBÍ AKTIVACI JISTIČE/POJISTKY NEBO
DOKONCE VZNÍCENÍ, POKUD SE JISTIČ/POJISTKA
NEUVEDE DO PROVOZU.
UPOZORNĚNÍ: V případě alarmu nebo
zjištění, že kyslíkový koncentrátor
nepracuje správně, nahlédněte do části
Odstraňování problémů v této příručce.
Pokud se vám nepodaří problém vyřešit,požádejte o radu dodavatele přístroje.
UPOZORNĚNÍ: Abyste zabránili zneplatnění záruky,dodržujte všechny pokyny výrobce.

VAROVÁNÍ: NEPOKOUŠEJTE SEO ŽÁDNOU ÚDRŽBU S VÝJIMKOU
MOŽNÝCH ŘEŠENÍ UVEDENÝCHV TÉTO PŘÍRUČCE.
VAROVÁNÍ: POUŽÍVEJTE POUZE PŘÍSLUŠENSTVÍDOPORUČENÉ VÝROBCEM. POUŽITÍ JAKÉHOKOLIJINÉHO PŘÍSLUŠENSTVÍ MŮŽE BÝT NEBEZPEČNÉA MŮŽE ZPŮSOBIT VÁŽNÉ POŠKOZENÍ VAŠEHOKYSLÍKOVÉHO KONCENTRÁTORU A POVEDEK ZNEPLATNĚNÍ ZÁRUKY.
VAROVÁNÍ: VÝROBCE DOPORUČUJE
ALTERNATIVNÍ ZDROJ DOPLŇKOVÉHO KYSLÍKU
PRO PŘÍPADY VÝPADKU PROUDU NEBO STAVU
ALARMU ČI MECHANICKÉ ZÁVADY PŘÍSTROJE.
O TYPU POŽADOVANÉHO ZÁLOŽNÍHO SYSTÉMUSE PORAĎTE SE SVÝM LÉKAŘEM NEBO
DODAVATELEM PŘÍSTROJE.
VAROVÁNÍ: ZABRAŇTE TOMU, ABY DOŠLOK ZABLOKOVÁNÍ PŘÍVODU VZDUCHU A VÝSTUPUVYFUKOVANÉHO VZDUCHU. NEUPUSŤTE ANINEVKLÁDEJTE ŽÁDNÉ PŘEDMĚTY DO OTVORŮ
PŘÍSTROJE. MOHLO BY DOJÍT K PŘEHŘÁTÍ
KYSLÍKOVÉHO KONCENTRÁTORU A SNÍŽENÍJEHO VÝKONU.
VAROVÁNÍ: KOUŘENÍ PŘI POUŽÍVÁNÍ KYSLÍKUJE HLAVNÍ PŘÍČINOU ZRANĚNÍ ZPŮSOBENÝCH
OHNĚM A S TÍM SOUVISEJÍCÍCH ÚMRTÍ. MUSÍTE
DODRŽOVAT NÁSLEDUJÍCÍ BEZPEČNOSTNÍ
VAROVÁNÍ:VAROVÁNÍ: VYVARUJTE SE KOUŘENÍ A POUŽITÍ
SVÍČEK ČI OTEVŘENÉHO OHNĚ VE STEJNÉMÍSTNOSTI, KDE SE NACHÁZÍ PŘÍSTROJ NEBO
PŘÍSLUŠENSTVÍ PŘENÁŠEJÍCÍ KYSLÍK.
VAROVÁNÍ: KOUŘENÍ PŘI NOŠENÍ KYSLÍKOVÉ
KANYLY MŮŽE ZPŮSOBIT POPÁLENINY OBLIČEJEA MOŽNÁ I SMRT.
VAROVÁNÍ: POKUD KOUŘÍTE, MOHOU VÁMNÁSLEDUJÍCÍ 3 KROKY ZACHRÁNIT ŽIVOT:VYPNĚTE KYSLÍKOVÝ KONCENTRÁTOR,SEJMĚTE KANYLU A OPUSŤTE MÍSTNOST,V NÍŽ SE TENTO PŘÍSTROJ NACHÁZÍ.

PN MN232-C4 L | Uživatelská příručka CZ - 451

FreeStyle Comfort
VAROVÁNÍ: V DOMÁCNOSTI NEBO V MÍSTECH,KDE SE POUŽÍVÁ KYSLÍK, MUSÍ BÝT NA
VIDITELNÉM MÍSTĚ UMÍSTĚNY NÁPISY „ZÁKAZ
KOUŘENÍ – POUŽÍVÁ SE KYSLÍK“. UŽIVATELÉA JEJICH PEČOVATELÉ MUSÍ BÝT INFORMOVÁNIO NEBEZPEČÍCH KOUŘENÍ V PŘÍTOMNOSTILÉKAŘSKÉHO KYSLÍKU NEBO PŘI JEHO POUŽITÍ.
VAROVÁNÍ: NEPOUŽÍVEJTE KYSLÍKOVÝ
KONCENTRÁTOR V PŘÍTOMNOSTI HOŘLAVÝCH
PLYNŮ. TO MŮŽE VÉST K RYCHLÉMU HOŘENÍ,KTERÉ ZPŮSOBÍ ŠKODY NA MAJETKU,TĚLESNÁ ZRANĚNÍ NEBO SMRT.
VAROVÁNÍ: ODSTRANĚNÍ KANYLY A JEJÍULOŽENÍ NA OBLEČENÍ, POVLEČENÍ, POHOVKUNEBO JINÝ VÝPLŇOVÝ MATERIÁL ZPŮSOBÍ PO
KONTAKTU S CIGARETOU, ZDROJEM TEPLANEBO PLAMENEM OKAMŽITÝ OHEŇ.
VAROVÁNÍ: NOSNÍ KANYLU NENECHÁVEJTENA OBLEČENÍ, POKRÝVCE POSTELE ČI
POLSTROVÁNÍ KŘESLA ANI POD NIMI. POKUDJE JEDNOTKA ZAPNUTÁ, ALE NEPOUŽÍVÁ SE,
KYSLÍK UČINÍ Z TĚCHTO MATERIÁLŮ HOŘLAVÉ
LÁTKY. ZAJISTĚTE, ABY BYL KYSLÍKOVÝ
KONCENTRÁTOR VYPNUTÝ, KDYŽ SE NEPOUŽÍVÁ.
VAROVÁNÍ: SCHVÁLENÉ PŘÍSLUŠENSTVÍ PRO
PODÁVÁNÍ KYSLÍKU JE NAVRŽENO TAK, ABYOMEZOVALO ŠÍŘENÍ POŽÁRU V ZÁJMU BEZPEČNOSTI PACIENTA. POUŽITÍ NESCHVÁLENÉHOPŘÍSLUŠENSTVÍ MŮŽE ZVÝŠIT RIZIKO ZRANĚNÍ
NEBO POŠKOZENÍ V SOUVISLOSTI S POŽÁREM.
VAROVÁNÍ: PŘED VYJMUTÍM AKUMULÁTORU
MUSÍ BÝT KYSLÍKOVÝ KONCENTRÁTORVYPNUTÝ A ODPOJENÝ OD VNĚJŠÍHO NAPÁJENÍ.
VAROVÁNÍ: MALÉ DÍLY MOHOUPŘEDSTAVOVAT POTENCIÁLNÍ RIZIKOUDUŠENÍ.
VAROVÁNÍ: VYVARUJTE SE KONTAKTU S OTVORYVYFUKOVANÉHO VZDUCHU NA DOBU DELŠÍ NEŽ
10 SEKUND, PROTOŽE MOHOU BÝT TEPLÉ.
VAROVÁNÍ: VÍTR ČI SILNÝ PRŮVAN MŮŽENEPŘÍZNIVĚ OVLIVNIT SPRÁVNÝ PRŮBĚH LÉČBY
KYSLÍKEM.
VAROVÁNÍ: TENTO VÝROBEK VÁS MŮŽEVYSTAVIT CHEMICKÝM LÁTKÁM, VČETNĚ NIKLU,KTERÝ JE VE STÁTĚ KALIFORNIE POVAŽOVÁNZA KARCINOGEN. PODROBNĚJŠÍ INFORMACE
NAJDETE NA WWW.P65WARNINGS.CA.GOV.
UPOZORNĚNÍ: Před odpojením zdroje
napájení vždy odpojte zdroj napájení zesítě a teprve poté odpojte zdroj napájení
od kyslíkového koncentrátoru.
452- CZ PN MN232-C4 L | Uživatelská příručka

UPOZORNĚNÍ: Neumisťujte kyslíkový koncentrátortak, aby byl obtížný přístup k napájecímu kabelu.
UPOZORNĚNÍ: Přívodní hadice kyslíku a napájecí
kabely vždy umístěte takovým způsobem,
aby se zabránilo riziku zakopnutí nebo možnému
náhodnému uškrcení.
UPOZORNĚNÍ: Při používání kyslíkového
koncentrátoru v automobilu, na lodi nebo u jiných
zdrojů stejnosměrného proudu v kombinaci
s napájecím zdrojem pro stejnosměrný proudpřed připojením kyslíkového koncentrátoru
zajistěte, aby bylo vozidlo nastartované a v chodu.
Nedodržení těchto pokynů může vést k tomu,
že napájecí zdroj nebude do kyslíkového
koncentrátoru dodávat energii.
UPOZORNĚNÍ: Když je automobil, ve kterémpoužíváte kyslíkový koncentrátor, vypnutý, přístroj
odpojte a vyndejte jej z automobilu. Neskladujte
kyslíkový koncentrátor ve velmi horkém nebo
chladném automobilu ani v žádném jiném
podobném prostředí s vysokými nebo nízkými
teplotami.
Poznámka: Přenosná a mobilní radiofrekvenční
komunikační zařízení mohou mít vliv na zdravotnické
elektrické přístroje.
UPOZORNĚNÍ: NENECHÁVEJTE
kyslíkový koncentrátor ani napájecí
zdroj připojený k vozidlu, pokud je
zapalování vypnuto. Pokud tak učiníte,může dojít k vybití baterie vozidla.
UPOZORNĚNÍ: Používání nebo skladování
kyslíkového koncentrátoru mimo rozsah jehoběžných provozních teplot a rozsah okolního tlaku(nadmořské výšky) může ovlivnit výkon a snížit
provozní dobu baterie nebo zvýšit dobu nabíjení
baterie. Mezní hodnoty skladovací a provozní
teploty najdete v části této příručky věnované
specifikacím.
UPOZORNĚNÍ: U kyslíkových koncentrátorů
vybavených bateriemi: Skladujte je na chladném
a suchém místě, abyste pomohli zajistit dlouhouživotnost baterie. Dlouhodobé skladování
kyslíkového koncentrátoru při vysokých teplotách
nebo s plně nabitou či zcela vybitou baterií můžesnížit celkovou životnost baterie. Nepokoušejte
se otevřít baterii – uvnitř baterie nejsou žádné
díly určené k servisu. Baterie uchovávejte mimo
dosah dětí.
UPOZORNĚNÍ: POUŽÍVEJTE POUZE baterie
dodané výrobcem. Pro řádnou likvidaci bateriepožádejte o pokyny dodavatele zařízení nebo místní
státní úřad.

FreeStyle Comfort
Poznámka: Pro zmírnění škod způsobených
koncentrátorem nebo výpadkem napájení by měl
poskytovatel zařízení posoudit potřebu záložního
kyslíku během instalace a pravidelně, když se tyto
atributy změní:Poznámka: Pokud byl kyslíkový koncentrátor delšídobu uskladněn mimo běžné provozní podmínkyznázorněné v tabulce specifikací, nechte jejpřed zapnutím zahřát/vychladnout na běžnou
provozní teplotu.

- Stav pacienta

Poznámka: Poskytovatel zařízení by měl posouditpotřebu záložního kyslíku pro případ výpadku
koncentrátoru nebo napájení při instalaci a pravidelně
při změně těchto atributů:- Prostředí, ve kterém pacient žije- Možnost doplnit pacientovi zásoby doplňkového
kyslíku.
Poznámka: Přesvědčte se, že je kanyla zcela
zasunutá a zajištěná. Díky tomu dokáže kyslíkový
koncentrátor správně rozeznat nadechnutí a dodávat
kyslík. Během nádechu byste měli slyšet nebo cítit,jak kyslík proudí do hrotů nosní kanyly.

- Stav pacienta- Prostředí, ve kterém pacient žije- Možnost doplnit pacientovi zásoby
doplňkového kyslíku.

Poznámka: Lithiové baterie mohou permanentně
ztrácet kapacitu, když jsou v plně nabitém nebo zcela
vybitém stavu vystaveny extrémně vysokým teplotám.
Za účelem dlouhodobého skladování se doporučuje,aby baterie byly nabité na 25–50 % a uchovávaly sev teplotním rozsahu –20 až +20 °C (–4 až +68 °F).
Poznámka: Kyslíkový koncentrátor nepoužívejte
bez nainstalovaného filtru na přívodu vzduchu.
Je-li k dispozici druhý filtr, vložte do přístroje náhradnífiltr před vyčištěním zaneseného filtru. Zanesený filtr
vyčistěte v teplé mýdlové vodě a před použitím jejřádně vysušte.
Poznámka: Výrobce nedoporučuje sterilizaci tohotozařízení.
Poznámka: Při použití kyslíkového koncentrátoru
nikdy nehrozí unikání kyslíku do místnosti.
VAROVÁNÍ: KYSLÍKOVÝ
KONCENTRÁTOR BY SE NEMĚL
POUŽÍVAT ANI PŘIPOJOVATK EXTERNÍMU ZDROJI STŘÍDAVÉHONEBO STEJNOSMĚRNÉHO PROUDU,POKUD NENÍ NAINSTALOVANÁBATERIE. JESTLIŽE JE BATERIEVYJMUTÁ, NEPOKOUŠEJTE SEJEDNOTKU ZAPNOUT ANI PŘIPOJIT
K EXTERNÍMU ZDROJI NAPÁJENÍ.
BATERII VYJMĚTE POUZE VE CHVÍLI, KDY JE
JEDNOTKA VYPNUTÁ A ODPOJENÁ OD PŘÍVODU
NAPÁJENÍ. NEPOKOUŠEJTE SE BATERIIVYJMOUT, KDYŽ JE JEDNOTKA STÁLE ZAPNUTÁ
NEBO PŘIPOJENÁ K PŘÍVODU NAPÁJENÍ.

PN MN232-C4 L | Uživatelská příručka CZ - 453

FreeStyle Comfort
Specifikace
FreeStyle ComfortRozměry (s jednou baterií)

10,0 x 7,3 x 3,1 in (25,4 x 18,5 x 7,9 cm)

Rozměry (s dvojitou baterií)

11,0 x 7,3 x 3,1 in (27,9 x 18,5 x 7,9 cm)

Hmotnost (s jednou baterií)

2,3 kg (5‫ ؘ‬lb)

Hmotnost (s dvojitou baterií)2,7 kg (6‫ ؘ‬lb)

Jmenovitá hladina hluku***

39,93 dB(A) ± 0,42 dB(A) @ 2

Nastavení průtoku

Pulzní nastavení 1, 2, 3, 4, 5

Nastavení výstupu O2

Nastavení 1 – 210 ml/min
Nastavení 2 – 420 ml/min
Nastavení 3 – 630 ml/min
Nastavení 4 – 840 ml/min
Nastavení 5 – 1050 ml/min([@20 bpm STPD] Všechny objemy +/- 15 %)Koncentrace kyslíku*
Monitor kyslíku
Maximální výstupní tlak
Citlivost dávkování
Napájení střídavým proudem
Napájení stejnosměrným proudem
Typ baterie
Kapacita baterie
Doba nabíjení baterieVýdrž baterií
Provozní teplota**

90 % (+5.5 % / -3 %)Ano< 206,8 kPa (30 psi)
> –0,1 cm H2O (nastavitelné)
100–240 V st, 50–60 Hz, 120 W Max
11–18 V ss (max. 10 A)
Lithium-iontové
Jednočlánková: 6700 mAh; Dvoučlánková: 13400 mAh
Jedna baterie: 3,5 hodiny
Dvojitá baterie: 6,0 hodinVýdrž jedné baterie: Nastavení 2 – až 4 hodinyVýdrž dvojité baterie: Nastavení 2 – až 8 hodin
5 až 40 °C (41 až 104 °F)Přepravní a skladovací teplota (Jednotka)–25 až +70 °C (–13 až +158 °F) do 3048 m (10 000 stop)Přepravní a skladovací teplota (Baterie)–25 až +70 °C (–13 až +158 °F) do 3048 m (10 000 stop)

Provozní nadmořská výška**Provozní vlhkost−381 až 3048 m (−1250 až 10 000 stop)(testováno na 700–1060 hPa)
15–95 % relativní vlhkosti (bez kondenzace)

Skladovací vlhkost

0–90% nekondenzující

Omezená záruka

Viz prohlášení o záruce* Na základě atmosférického tlaku 101 kPa (14,7 psi) při 21 °C (70 °F)** Jednotka bude fungovat ve všech prostředích, pokud je prostředí se specifikovanými teplotními a vlhkostními rozsahy.
Provoz mimo tyto provozní specifikace může omezit schopnost kyslíkového koncentrátoru splnit uvedené parametryu vyšších průtočných množství. Obrazovka rozehřívání se může nadále zobrazovat, dokud koncentrátor nedosáhne cílové
koncentrace nebo po dobu až 15 minut. Během této doby jednotku nadále používejte.
*** Hladina zvuku měřená podle zkušební metody č. 14-1 10/2018 MDS-Hi.
Předpokládaná životnost je nejméně pět let.

454- CZ PN MN232-C4 L | Uživatelská příručka

FreeStyle Comfort
Provozní pokyny
1. Umístěte přístroj FreeStyle Comfort
na vhodné místo.
a. Koncentrátor by měl být dobře větrán – filtryna přívodu vzduchu a výstup vzduchu byneměly být nijak blokovány.
b. Koncentrátor by měl být umístěn takovým
způsobem, aby všechny zvukové a vizuální
ukazatele a výstrahy byly snadno slyšeta vidět.
c. Před použitím přístroje se přesvědčte,že jsou nainstalovány filtry na přívodu
vzduchu. Pokud přístroj nemá filtr,
kontaktujte dodavatele zařízení.

3. Zasuňte konektor napájecího zdroje pro střídavý
nebo stejnosměrný proud do samice napájecího
konektoru v pravém dolním rohu přední části
přístroje.

2. Nainstalujte baterii do koncentrátoru.
a. Zarovnejte baterii se spodní středovou částí
krytu přístroje FreeStyle Comfort.

4. Připojte nosní kanylu k výstupu kyslíku. Předpoužitím se přesvědčte, že zapojení je zajištěné.
POZNÁMKA: Použití jakýchkoli kanyl, které nebyly
doporučeny, může ovlivnit výkon.

Poznámka: Pro dodavatele zařízení: K použití
s FreeStyle Comfort se doporučuje následujícípříslušenství pro vedení kyslíku:b. Zasuňte baterii dovnitř, dokud neuslyšítezacvaknutí, které potvrdí zajištění baterie na
správném místě.

• Nosní kanyla s hadičkou dlouhou 2,1 m (7 stop)(maximální l/min: 6 l/min): Číslo dílu CAIRE 5408-SEQ• Další doporučené informace o příslušenství jsou
k dispozici on-line na adrese www.caireinc.com.
5. Stisknutím tlačítka vypínače zapněte přístroj
FreeStyle Comfort a vyberte předepsané
nastavení průtoku pomocí tlačítek ke zvýšení (+)
nebo snížení (–) průtoku.

PN MN232-C4 L | Uživatelská příručka CZ - 455

FreeStyle Comfort
Frekvencedýchání
15
20
25
30
35
40

Uslyšíte krátkou zvukovou výstrahu, která
signalizuje, že došlo k zapnutí přístroje.
Poznámka: Při prvním zapnutí přístroje FreeStyle
Comfort je průtok nastaven na hodnotu 2. Pokud
dojde ke změně nastavení, přístroj se příště zapne
s posledním nastavením průtoku.
6. Nasaďte si nosní kanylu na obličej dle pokynů
výrobce dodaných s kanylou a dýchejte běžným
způsobem. Při každém nadechnutí byste mělicítit dávku kyslíku vycházející z nosní kanyly.
7. Nastavení průtoku na požadované průtočnémnožství změníte pomocí tlačítek pro zvýšení
(+) nebo snížení (–) průtoku. Při změně
nastavení průtoku je běžné, že uslyšíte rozdílné
zvuky.

Frekvencedýchání
15
20
25
30
35
40

Nastavení pulzního objemu autoSAT (ml)*
1
2
3
4
5
10,5
10,5
10,5
10,5
10,5
10,5

21,0
21,0
21,0
21,0
21,0
21,0

31,5
31,5
31,5
31,5
30,0
26,3

42,0
42,0
42,0
35,0
30,0
26,3

Tolerance je ±15 %456- CZ PN MN232-C4 L | Uživatelská příručka

52,5
52,5
42,0
35,0
30,0
26,3

1
14,0
10,5
8,4
7,0
6,0
5,3

Nastavení pulzního objemu (ml) –nikoli autoSAT
2
3
4
5
28,0
21,0
16,8
14,0
12,0
10,5

42,0
31,5
25,2
21,0
18,0
15,8

56,0
42,0
33,6
28,0
24,0
21,0

70,0
52,5
42,0
35,0
30,0
26,3

8. Vypněte přístroj FreeStyle Comfort stisknutím
tlačítka vypínače.
Poznámka: Obrazovka displeje přejde do
pohotovostního režimu a po 20 sekundách zhasne.
Obrazovku probudíte z pohotovostního režimu
stisknutím kteréhokoli tlačítka.
9. Pokud jde o pokročilé funkce, viz Pokyny pro
navigaci v menu.

SPECIÁLNÍ FUNKCE PRODUKTU
autoSAT®
Patentovaná funkce autoSAT společnosti CAIREumožňuje zařízení FreeStyle Comfort dodávat
konstantní objem kyslíku (bolus) na jeden nádech
až do maximální dechové frekvence. Vzhledem
k tomu, že se dechová frekvence uživatele zvyšujeběhem aktivit, jako je chůze, funkce autoSAT
upravuje provoz zařízení tak, aby byl zachován
nastavený objem bolusu.
UltraSense®
Tato citlivá spoušť pulzní dávky detekuje změnu
tlaku během vašeho dýchání a uvolňuje proudkyslíku, když se nadechnete.
autoDOSE®
Pokud dojde ke spuštění poplachu „nebyl zjištěn
nádech“, tato funkce spustí proud kyslíku při
20 nádeších za minutu se stávajícím nastavenímprůtoku, dokud nebude stav poplachu vyřešen.

FreeStyle Comfort
Napájecí zdrojePřístroj FreeStyle Comfort lze napájettřemi způsoby:• Nabíjecí baterie• Napájení střídavým proudem• Napájení stejnosměrným proudem
Informace o provozu všech těchto součástí jsou
uvedeny v následujících odstavcích.

VYJMUTÍ BATERIE

Stiskněte tlačítko pro vysunutí baterie dolů směrem
od přístroje FreeStyle Comfort. DRŽTE tlačítko
a vytáhněte baterii směrem od koncentrátoru.
VAROVÁNÍ: PŘED VYJMUTÍM BATERIEMUSÍ BÝT KYSLÍKOVÝ KONCENTRÁTORVYPNUTÝ A ODPOJENÝ OD EXTERNÍHO
NAPÁJENÍ.

VAROVÁNÍ: POUŽÍVEJTE POUZENAPÁJECÍ ZDROJE UVEDENÉ V TÉTOUŽIVATELSKÉ PŘÍRUČCE. POUŽITÍ
NAPÁJECÍCH ZDROJŮ NEBO BATERIÍ,KTERÉ NEPOCHÁZÍ OD SPOLEČNOSTICAIRE, MŮŽE ZPŮSOBIT POŠKOZENÍ
PŘÍSTROJE A JEHO NESPRÁVNÝ
PROVOZ.

BATERIE

Přístroj FreeStyle Comfort lze používat s dobíjecí
baterií, která je dodávána s přístrojem. Při použití
kyslíkového koncentrátoru s baterií neexistujížádná provozní omezení. Při napájení baterií jsou
dostupná všechna průtočná množství.

KONTROLA STAVU NABITÍ BATERIEKdyž je baterie nainstalována v přístroji FreeStyle
Comfort, zjistíte její stav nabití na displeji pomocí
ikony baterie a odhadované zbývající doby.
80–100 %INSTALACE BATERIE

Zarovnejte baterii se spodní středovou částí krytu
přístroje FreeStyle Comfort. Zatlačte baterii
dovnitř, dokud neuslyšíte zacvaknutí, které potvrdí
zajištění baterie na správném místě.

60–80 %

40–60 %

20–40 %20 %PN MN232-C4 L | Uživatelská příručka CZ - 457

FreeStyle Comfort
Poznámka: Pokud je připojen napájecí zdroj, zobrazí
se místo zbývajících procent nabití indikátor zástrčky.
Chcete-li zjistit stav nabití u baterie, která nenínainstalována v přístroji FreeStyle Comfort, položte
ji a stiskněte tlačítko TEST. Rozsvítí se signalizační
kontrolky baterie (25–100 %) umístěné vpravo
od tlačítka TEST, které informují o úrovni nabitíbaterie.

PROVOZNÍ VÝDRŽ BATERIE

Je-li externí napájecí zdroj pro střídavý nebo
stejnosměrný proud odpojen a je nainstalovaná
nabitá baterie, koncentrátor se automaticky přepne
na napájení z baterie. Níže uvedená tabulka ukazuje
typickou výdrž nové baterie. Výdrž baterie zůstaneneměnná bez ohledu na vaši frekvenci dýchání.
Nastavenípulzní
dávkyVýdrž jedné(8článkové)baterieVýdrž dvojité(16článkové)baterie1

až 8 hodinaž 16 hodin2

až 4 hodiny

až 8 hodin3

až 3 hodinyaž 6 hodin4až 2,25 hodiny

až 4,5 hodiny

5až 2 hodiny

až 4 hodiny

TYPICKÁ DOBA DOBÍJENÍ BATERIEPRVNÍ DOBÍJENÍ

Nová baterie dodávaná s přístrojem FreeStyle
Comfort je při expedici z výrobního závodu nabita
pouze částečně. Před prvním použitím byste mělibaterii dobít pomocí napájecího zdroje pro střídavý
nebo stejnosměrný proud.

OBECNÉ INFORMACE O DOBÍJENÍ

Přístroj FreeStyle Comfort bude dobíjet bateriipokaždé, když je nainstalována do koncentrátoru
a ten je připojen k externímu napájecímu zdroji prostřídavý nebo stejnosměrný proud (kromě letadel).
Baterie je plně nabitá, když ukazuje stav 100 %.
Baterii lze také dobíjet pomocí volitelné stolnínabíječky.

458- CZ PN MN232-C4 L | Uživatelská příručka

Typická doba plného dobití zcela vybité baterieje 3,5 hodiny u jedné baterie a 6 hodin u dvojitébaterie.
Poznámka: Pokud byla baterie zcela vybitá, nemusí
se začít dobíjet ihned.

ŽIVOTNOST BATERIE

Baterie přístroje FreeStyle Comfort jsou navrženy
tak, aby vydržely 300 cyklů nabití/vybití.

FreeStyle Comfort
Napájecí zdroj pro střídavý
a stejnosměrný proudPřístroj FreeStyle Comfort je standardně dodáváns napájecími zdroji pro střídavý i stejnosměrný proud.
Napájecí zdroj pro střídavý proud obsahuje
následující položky:	a. Napájecí adaptér pro střídavý proud, který
pracuje s napětím 100–240 V při frekvenci
50–60 Hz.
	b. Napájecí kabel pro střídavý proud určený
k připojení k napájecímu adaptéru a příslušné
zásuvce se střídavým proudem.

Napájecí zdroj pro stejnosměrný proud se skládá
z kabelu, který se zapojí přímo do jednotky a do
zásuvky se stejnosměrným proudem.

2. Připojte napájecí kabel pro střídavý proudk napájecímu adaptéru pro střídavý proud.

NAPÁJENÍ STŘÍDAVÝM PROUDEM

Napájení střídavým proudem je určeno propoužití doma nebo na místech, kde je k dispozici
standardní zásuvka se střídavým proudem.
Přístroj FreeStyle Comfort je v kombinaci
s napájením střídavým proudem plně funkční.
Přístroj FreeStyle Comfort také při každém
připojení k napájení střídavým proudem dobíjí
baterii (kromě letadel). Baterie se dobíjí bez ohledu
na to, zda je přístroj FreeStyle Comfort zapnutý
nebo vypnutý.

3. Opačný konec napájecího kabelu pro střídavý
proud zapojte do zásuvky ve zdi nebo jinévhodné elektrické zásuvky.

VAROVÁNÍ: NEPŘIPOJUJTE EXTERNÍ
NAPÁJENÍ, POKUD NENÍ BATERIENAINSTALOVANÁ V KYSLÍKOVÉM
KONCENTRÁTORU.
Přístroj FreeStyle Comfort připojíte k napájení
střídavým proudem následujícím způsobem:
1. Zapojte napájecí adaptér pro střídavý proud donapájecí zásuvky v pravém dolním rohu přední
části přístroje FreeStyle Comfort. Dejte pozor,
abyste konektor zatlačili až nadoraz a připojení
tak bylo zajištěné.

PN MN232-C4 L | Uživatelská příručka CZ - 459

FreeStyle Comfort
4. Stisknutím tlačítka vypínače zapněte přístroj
FreeStyle Comfort a vyberte předepsané
nastavení průtoku pomocí tlačítek ke zvýšení (+)
nebo snížení (–) průtoku.

4. Stisknutím tlačítka vypínače zapněte přístroj
FreeStyle Comfort a vyberte předepsané
nastavení průtoku pomocí tlačítek ke zvýšení (+)
nebo snížení (–) průtoku.
Poznámka: Pokud je potřeba dobít baterii, ikona bateriena displeji bude mít jeden oranžový sloupeček.

NAPÁJENÍ STEJNOSMĚRNÝM PROUDEM

Napájení stejnosměrným proudem je určeno propoužití se zásuvkami v automobilech, na lodích nebov jiných motorových vozidlech. Přístroj FreeStyle
Comfort je v kombinaci s napájením stejnosměrným
proudem plně funkční. V případě potřeby přístroj
FreeStyle Comfort také při každém připojení
k napájení stejnosměrným proudem dobíjí baterii(kromě letadel). Baterie se dobíjí bez ohledu na to,zda je přístroj vypnutý nebo zapnutý a používá se.
VAROVÁNÍ: NEPŘIPOJUJTEEXTERNÍ NAPÁJENÍ, POKUD NENÍV KONCENTRÁTORU KYSLÍKUNAINSTALOVÁN AKUMULÁTOR.
1. Nastartujte motor vozidla.
2. Zapojte konektor napájecího zdroje pro
stejnosměrný proud do samice napájecího
konektoru v pravém dolním rohu přední části
přístroje FreeStyle Comfort. Dejte pozor, abyste
konektor zatlačili až nadoraz a připojení tak bylo
zajištěné.

3. Opačný konec napájecího kabelu pro
stejnosměrný proud zapojte do zásuvky
automobilu se stejnosměrným proudem.
460- CZ PN MN232-C4 L | Uživatelská příručka

UPOZORNĚNÍ: Abyste předešli možnému
požáru nebo přehřátí, ujistěte se, že v
zásuvce stejnoměrného proudu o napětí
12V nejsou žádné nečistoty a že zástrčka
adaptéru stejnosměrného napájecího
zdroje správně zapadá a zůstává pevně
na svém místě.
UPOZORNĚNÍ: Kovové povrchy stejnosměrného
napájení mohou být při používání VELMI HORKÉz důvodu vysokého proudu. Před dotykem nechte
povrchy dostatečně dlouho vychladnout.
UPOZORNĚNÍ: Nepoužívejte stejnosměrný napájecí
zdroj s žádnými prodlužovacími prvky, rozbočovači
nebo adaptéry. Použití neschváleného příslušenstvímůže vést k jeho přehřátí nebo poškození.
UPOZORNĚNÍ: Připojení zařízení k elektrické
zásuvce 12V DC může vyčerpat baterii vozidla,
pokud motor neběží, a to i v případě, že je zařízení
vypnuté.
UPOZORNĚNÍ: Vozidlo nestartujte, když je připojen
stejnosměrný napájecí zdroj.

FreeStyle ComfortČištění, péče a pravidelná údržba
PRAVIDELNÁ ÚDRŽBAPreventivní údržba
Zařízení je bezúdržbové. Filtry uvnitř zařízení
jsou určeny k tomu, aby vydržely po celou
dobu životnosti zařízení. Pokud zařízení
nefunguje správně, jednotka by měla být vrácena
poskytovateli k opravě. Servis vnitřních součástí
umístěných pod krytem přístroje FreeStyle Comfort
smí provádět pouze dodavatel kyslíku.
Obsluha jednotky je zodpovědná za následující
postup údržby.
• Jednou týdně – vyčistěte filtry na přívodu vzduchu
VAROVÁNÍ: NEPOKOUŠEJTE SE OTEVŘÍT ANI SEJMOUT PLASTOVÉ KRYTY
KYSLÍKOVÉHO KONCENTRÁTORU.
PŘÍSTROJ SMÍ OTVÍRAT POUZE KVALIFIKOVANÝ SERVISNÍ TECHNIK.

FILTRY NA PŘÍVODU VZDUCHU

Vzduch vstupuje do přístroje FreeStyle Comfort přesfiltry na přívodu vzduchu umístěné v přední spodníčásti kyslíkového koncentrátoru. Tyto filtry zabraňují
tomu, aby do přístroje pronikal prach nebo jiné velkéčástice, které se nachází ve vzduchu.
Filtry na přívodu vzduchu by se měly čistit alespoň
jednou týdně. Dodavatel kyslíku vám můžev závislosti na provozních podmínkách doporučitčastější čištění filtrů. Za účelem řádného čištění filtrů
na přívodu vzduchu dodržujte následující kroky.
1. Vyjměte oba filtry z přední části přístroje
FreeStyle Comfort zatažením za příslušný
jazýček a jejich vytažením směrem ven.

VAROVÁNÍ: NEPOUŽÍVEJTEKYSLÍKOVÝ KONCENTRÁTOR BEZ
NAINSTALOVANÉHO FILTRU NAPŘÍVODU VZDUCHU.

ÚDRŽBA BATERIEChraňte baterii před kontaktem s kapalinami.
Nepokoušejte se baterii rozebírat ani otevírat.
Nevystavujte baterii otevřeným plamenům.
Nepouštějte baterii na zem.
Uchovávejte baterii mimo dosah dětí.
Pokud baterie vykazuje známky poškození, předpoužitím kontaktujte dodavatele kyslíkového
koncentrátoru.
Nepokládejte baterii do mikrovlnné trouby či jiných
kuchyňských zařízení ani do jejich blízkosti.

SKLADOVÁNÍ BATERIE

Baterie kyslíkového koncentrátoru nemusí být předdobíjením zcela vybitá. Doporučuje se dobíjet baterii
kyslíkového koncentrátoru po každém použití.
Nenechávejte baterii na místech vystavených
nadměrnému teplu, například na přímém slunečnímsvětle nebo v automobilu. Pokud je na takovémmístě necháte, může to způsobit snížení výkonu
baterie a zkrácení její životnosti. Může to vést
také k tomu, že baterie bude produkovat teplo
či kouř, dojde k jejímu vznícení nebo vybuchne.
Baterie by měla být skladována ve stabilním
prostředí s nízkou vlhkostí (nižší než 70 % relativnívlhkosti), ve kterém se nevyskytují korozivní plyny.
Abyste zabránili vzniku rzi, předejděte situacím,
při kterých může dojít ke srážení vody, například
rychlému střídání okolní teploty.
Za účelem dlouhodobého skladování (jeden měsíc
a déle) se doporučuje stav částečného nabití/vybitímezi 50 % až 75 % (dvě až tři kontrolky LED).
Jakmile je baterie nabitá, vyjměte ji z kyslíkového
koncentrátoru.

2. Umyjte oba filtry v teplé mýdlové vodě.
3. Ř
 ádně filtry opláchněte a zbývající vodu otřete
měkkým hadříkem bez žmolků. Přesvědčte se,že filtry jsou před opětovnou instalací zcela suché.
4. Nainstalujte zpět čisté a suché filtry.

PN MN232-C4 L | Uživatelská příručka CZ - 461

FreeStyle ComfortČIŠTĚNÍ

Před jakýmkoli čištěním přístroj VYPNĚTE
a odpojte od napájení střídavým či stejnosměrným
proudem. Na vnější kryt přímo NESTŘÍKEJTE.
Používejte vlhký (nikoli mokrý) hadřík nebo
houbičku. Pro vyčištění krytu a napájecích zdrojů
postříkejte hadřík nebo houbičku roztokem
jemného čisticího prostředku pro použití
v domácnosti. Postupujte dle pokynů výrobcečisticího prostředku.
VAROVÁNÍ: PŘED ČIŠTĚNÍM PŘÍSTROJVYPNĚTE A ODPOJTE NAPÁJECÍKABEL ZE SÍŤOVÉ ZÁSUVKY, ABYSTEZABRÁNILI NEBEZPEČÍ NÁHODNÉHOÚRAZU ELEKTRICKÝM PROUDEM
A VZNIKU POPÁLENIN. SNÍMATKRYTY A PROVÁDĚT SERVIS TOHOTO
PŘÍSTROJE SMÍ POUZE DODAVATELZAŘÍZENÍ NEBO KVALIFIKOVANÝSERVISNÍ TECHNIK.
VAROVÁNÍ: JE NUTNÉ DBÁT NA TO, ABY SEKYSLÍKOVÝ KONCENTRÁTOR A BATERIE
NENAMOČILI A ABY SE DO NICH NEDOSTALAŽÁDNÁ KAPALINA. MOHLO BY TO VÉSTK PORUŠE NEBO VYNUCENÉMU VYPNUTÍA ZVÝŠIT NEBEZPEČÍ ÚRAZU ELEKTRICKÝMPROUDEM NEBO VZNIKU POPÁLENIN.
VAROVÁNÍ: NEPOUŽÍVEJTE KAPALINY PŘÍMO
NA PŘÍSTROJ. SEZNAM NEŽÁDOUCÍCHCHEMICKÝCH LÁTEK OBSAHUJE MIMOJINÉ NÁSLEDUJÍCÍ POLOŽKY: ALKOHOLA VÝROBKY NA BÁZI ALKOHOLU, VÝROBKY
NA BÁZI KONCENTROVANÉHO CHLORU(ETHYLENDICHLORID) A VÝROBKY NABÁZI OLEJE. TYTO LÁTKY SE NESMÍ
POUŽÍVAT K ČIŠTĚNÍ PLASTOVÉHO KRYTUKONCENTRÁTORU, PROTOŽE MOHOU PŘÍSTROJPOŠKODIT.
VAROVÁNÍ: KRYT, OVLÁDACÍ PANEL A NAPÁJECÍKABEL OČISTĚTE POUZE POMOCÍ JEMNÉHODOMÁCÍHO ČISTICÍHO PROSTŘEDKU NANESENÉHO VLHKÝM (NIKOLI MOKRÝM) HADŘÍKEM NEBO
HOUBIČKOU A NÁSLEDNĚ VŠECHNY POVRCHYVYTŘETE DOSUCHA. ZABRAŇTE VNIKNUTÍ JAKÉKOLI KAPALINY DOVNITŘ KONCENTRÁTORU.
VĚNUJTE ZVLÁŠTNÍ POZORNOST VÝSTUPU KYSLÍKU URČENÉMU K PŘIPOJENÍ KANYLY, UJISTĚTE
SE, ŽE NA NĚM NENÍ ZACHYCEN PRACH, VODAANI JINÉ ČÁSTICE.

462- CZ PN MN232-C4 L | Uživatelská příručka

VAROVÁNÍ: K ČIŠTĚNÍ BATERIE POUŽIJTE VLHKÝ(NIKOLI MOKRÝ) HADŘÍK NEBO HOUBIČKU.
NEJDŘÍVE NASTŘÍKEJTE NA HADŘÍK NEBO
HOUBIČKU JEMNÝ DOMÁCÍ ČISTICÍ PROSTŘEDEKA OČISTĚTE KRYT BATERIE.
VAROVÁNÍ: K ČIŠTĚNÍ NEPOUŽÍVEJTE
AUTOMATIZOVANÉ MYCÍ SYSTÉMY.

VÝMĚNA KANYLYVždy postupujte podle pokynů ke správnémupoužití od výrobce kanyl. Nosní kanylu nebo
hadici vedoucí kyslík vyměňte po každých 3 až
6 měsících (při denním použití po 3 měsících)podle doporučení výrobce nebo dodavatele kyslíku.
Pokyny k čištění a výměně vám poskytne lékař
nebo dodavatel kyslíku.
Další produkty pro případ výměny jsou dostupné
u dodavatele kyslíku.

FreeStyle Comfort
Stavy alarmů
Priorita č.

Popisalarmu1

Poruchasystému,vysoká
teplotabaterie

Zobrazení na displeji

Zvukový
signálPodrobnosti / navrhované opatření

1 pípnutíkaždých
20 sekundKdyž se teplota baterie blíží mezníhodnotě bezpečnostního vypnutí, přístroj
vydá výstrahu.
Když je dosaženo mezní hodnoty vypnutí,přístroj se vypne. Je-li připojeno externínapájení, objeví se obrazovka výstrahy
a po dobu 2 minut se bude ozývat
zvuková výstraha.
Testujte s baterií při pokojové teplotě. Pokud
problém přetrvává, kontaktujte dodavatele.

2

Poruchasystému,vysoká
teplota1 pípnutíkaždých
20 sekundPřístroj vydá výstrahu, když se vnitřní
teplota blíží mezní hodnotě vypnutí.
Když je dosaženo mezní hodnoty vypnutí,přístroj se vypne, objeví se obrazovkavýstrahy a po dobu 2 minut se budeozývat zvuková výstraha.
Ujistěte se, že žádný otvor jednotky nenízablokovaný. Pokud problém přetrvává,
kontaktujte dodavatele.

3

Poruchasystému,vysoký tlak
(vysoký tlakv systému
nebo porucha
kompresoru)

1 pípnutíkaždých
20 sekund

4

Poruchasystému,průtočné
množství(porucha
ventilu OCD,překážka
v rozvodu
plynu)

1 pípnutíkaždých
30 sekund

Pokud problém přetrvává, kontaktujte
dodavatele.

5

Vypnutí kvůli
vybité baterii

1 pípnutíkaždých
30 sekundVýstraha je aktivní, dokud nedojdek úplnému vybití baterie a vypnutípřístroje.

Když je dosaženo mezní hodnoty vypnutí,přístroj se vypne, objeví se obrazovka výstrahy a po dobu 2 minut se bude ozývat
zvuková výstraha.
Pokud problém přetrvává, kontaktujte
dodavatele.

Nabijte baterii.

PN MN232-C4 L | Uživatelská příručka CZ - 463

FreeStyle Comfort
Priorita č.

Popisalarmu6

Nebyl zjištěn
dech

Zobrazení na displeji

Zvukový
signálPodrobnosti

1 pípnutíkaždých
10 minut

Po dobu 15 sekund nebyl zjištěn nádech.
Aktivuje se po prvním nádechu nebo pozhasnutí obrazovky rozehřívání (podle
toho, co nastane dříve).
Zkontrolujte, zda kanyla není zalomená
nebo zda na ní nejsou netěsnosti.
Zvyšte nastavení citlivosti. Pokud problémpřetrvává, kontaktujte dodavatele.

7ZahřívánížádnýPřístroj zahájí provoz automatickým
dávkováním, dokud nezjistí nádech
nebo neskončí rozehřívání (podle toho,co nastane dříve).

8

Nízká
koncentrace

1 pípnutíkaždých
10 minut,až
30 minut

Pokud problém přetrvává, kontaktujte
dodavatele.

9

Varování
vybitá baterie

1 pípnutí,
bez opakováníNabijte baterii.

10Překročená
frekvencedýchání1 pípnutí,
bez opakování

Frekvence dýchání překročila hodnotu
40 nádechů za minutu.

11

Porucha systému, závada
ventilátoru

žádný

Pokud problém přetrvává, kontaktujte
dodavatele.

12

Poruchasystému,nízký tlakžádnýPokud problém přetrvává, kontaktujte
dodavatele.

13

Porucha systému, nízká
teplotažádný

Nechte jednotku zahřát na pokojovou teplotu. Pokud problém přetrvává, kontaktujte
dodavatele.

464- CZ PN MN232-C4 L | Uživatelská příručka

Jestliže je dávka nesprávně spouštěna,snižte nastavení citlivosti. Pokud problémpřetrvává, kontaktujte dodavatele.

FreeStyle Comfort
Priorita č.

PopisalarmuInformace

Spuštění

Zobrazení na displeji

Zvukový
signálPodrobnosti

1 informační
pípnutíObrazovka se objeví při zapnutí. Mělibyste slyšet jedno pípnutí, abyste sepřesvědčili, že bzučák funguje.
Obrazovka spuštění ukazuje v pravém
spodním rohu aktuální verzi firmwaru.

Informace

Obrazovkaběžného chodu v režimu
autoSAT (nenípřipojen žádný
napájecí zdroj
stejnosměrného proudu)

žádný

Informace

Obrazovkaběžnéhochodu s deaktivovaným
autoSAT (nenípřipojen žádný
napájecí zdroj
stejnosměrného proudu)

žádný

InformaceZměnanastavení

1 informační
pípnutíInformacePřipojeníexterního
napájení vevypnutém
stavu

žádný

InformacePřipojeníexterního
napájení za
provozužádný

Informace

Odpojeníexterního
napájení vevypnutémstavu

1 informační
pípnutíNa 2 sekundy se objeví obrazovka
s logem společnosti CAIRE, potom se na
5 sekund zobrazí stav baterie a připojení
k napájení a následně obrazovka zhasne.

Na 2 sekundy se objeví obrazovka
s logem společnosti CAIRE a potom
se displej vypne.

PN MN232-C4 L | Uživatelská příručka CZ - 465

FreeStyle Comfort
Priorita č.

PopisalarmuZobrazení na displeji

Informace

Odpojeníexterního
napájeníza provozu(s nainstalovanou baterií)

1 informační
pípnutíInformace

V pohotovostním režimu
při stisknutí
tlačítka „nahoru“ nebo„dolů“ se na
displeji zobrazí stav bateriea nabíjení

žádný

InformaceRežimLetadlo při
napájeníz baterie

1 dlouhépípnutíInformaceRežimLetadlo při
zapojení
do sítě

1 dlouhépípnutí466- CZ PN MN232-C4 L | Uživatelská příručka

Zvukový
signálPodrobnosti

Obrazovka se objeví na 5 sekund a potom
zhasne.

FreeStyle Comfort
Pokyny pro navigaci v menu
Do menu můžete vstoupit v pohotovostním režimu
nebo v provozním režimu. V provozním režimu,jestliže nastane stav alarmu, bude menu automaticky
opuštěno. Jestliže po dobu 20 sekund nedojde ke
stisknutí žádného tlačítka, menu bude automaticky
opuštěno. V pohotovostním režimu budete možná
muset stisknout tlačítko Menu dvakrát, abyste
jednotku probudili, pak vstoupíte do menu.
Existují čtyři obrazovky menu:Režim Letadlo

1. Obrazovka stavu: Zobrazuje číslo rev. firmwaru,
dobu provozu, cykly vybití baterie
2. Režim Letadlo (ZAP/VYP)
3. Citlivost: nastavitelná (1-5)
4. autoSAT (ZAP/VYP)

CitlivostPříklad stavové obrazovkyOVLÁDÁNÍ MENU NASTAVENÍ

autoSATDo menu Nastavení vstoupíte stisknutím
tlačítka Menu/Zvolit
pohotovostním režimu.

v provozním nebo

Pro procházení nabídkami posunujte.

Stavová stránka

PN MN232-C4 L | Uživatelská příručka CZ - 467

FreeStyle Comfort
STAVOVÁ STRÁNKA
Stavová stránka FreeStyle Comfort zobrazuje revizi
firmwaru, provozní hodiny a počet cyklů baterie napřipojené baterii.

Chcete-li zrušit zapnutí režimu Letadlo, stiskněte
, když je vidět ikona CANCEL (ZRUŠIT).

Zapnutí režimu Letadlo potvrzeno

Stavová stránka

Provozní hodiny jsou uvedeny v celých hodinách.
Vybíjení baterie je počet cyklů, kdy byla připojená
baterie vybita a dobita.

REŽIM LETADLOFreeStyle Comfort se expeduje s vypnutým
režimem Letadlo.

Chcete-li režim Letadlo vypnout, opakujte výšeuvedené, zatímco přepnete na OFF (VYP.).
Stisknutí, když obrazovka ukazuje symbol„Next“ (Další), posune stránku menu k další
volitelné možnosti obrazovky.
Režim Letadlo je určen k použití v provoznímrežimu. Režim Letadlo bude automaticky
deaktivován, jestliže bude jednotka ponechána
v pohotovostním režimu při napájení z baterie.
Poznámka: Podrobnější informace o režimu Letadlo
najdete v části Aktivace režimu Letadlo.

REŽIM CITLIVOST

Hlavní obrazovka režimu Letadlo

Chcete-li tento režim zapnout, stiskněte
,když je vidět ikona CHANGE (ZMĚNIT).
Hodnota se přepne na modré slovo „ON“ (ZAP.).

FreeStyle Comfort se expeduje s výchozím
nastavením z výrobního závodu pro režim Citlivostv hodnotě 2 a za normálních podmínek by neměl
vyžadovat úpravu. Nastavení citlivosti lze seřizovat
mezi hodnotami 1–5, kde 1 je nejnižší citlivosta 5 je nejvyšší citlivost.

Obrazovka režimu Citlivost při nastavení 2
Je zvoleno Režim Letadlo ON (zapnutý)Pro potvrzení zapnutí režimu Letadlo stiskněte
, když je vidět ikona SAVE (ULOŽIT).
Hodnota se přepne na černé slovo „ON“ (ZAP.).

468- CZ PN MN232-C4 L | Uživatelská příručka

FreeStyle Comfort
Chcete-li upravit režim Citlivost, stiskněte
když je vidět ikona CHANGE (ZMĚNIT).
Hodnota se přepne na další modrou číslici.

,

Je zvoleno nastavení 3 režimu Citlivost

Pro potvrzení nastavení 3 v režimu Citlivost
stiskněte
, když je vidět ikona SAVE (ULOŽIT).
Hodnota se přepne na černou hodnotu „3“.

REŽIM AUTOSATFreeStyle Comfort se expeduje se zapnutým
režimem autoSAT.

Hlavní obrazovka režimu autoSAT

Chcete-li tento režim vypnout, stiskněte
,když je vidět ikona CHANGE (ZMĚNIT).
Hodnota se přepne na modré slovo „OFF“ (VYP.).

Chcete-li zrušit změnu režimu Citlivost, stiskněte
, když je vidět ikona CANCEL (ZRUŠIT).

Je zvolen Režim autoSAT Off (VYP)

Pro potvrzení vypnutí režimu autoSAT stiskněte
Nastavení 3 režimu Citlivost potvrzeno, když je vidět ikona SAVE (ULOŽIT).
Hodnota se přepne na černé slovo „OFF“ (VYP.).
Chcete-li zrušit zapnutí režimu autoSAT, stiskněte
, když je vidět ikona CANCEL (ZRUŠIT).

Nastavení 5 (max.) režimu Citlivost potvrzeno

Stisknutí, když obrazovka ukazuje symbol„Next“ (Další), posune stránku menu k další
volitelné možnosti obrazovky.

Vypnutí režimu autoSAT potvrzeno

Chcete-li režim autoSAT zapnout, opakujte výšeuvedené, zatímco přepnete na ON (ZAP.).
Stisknutí, když obrazovka ukazuje symbol„Next“ (Další), posune stránku menu k další
volitelné možnosti obrazovky.

PN MN232-C4 L | Uživatelská příručka CZ - 469

FreeStyle Comfort
Odstraňování problémů
Problém

Pravděpodobná příčina

Řešení

Přístroj FreeStyle Comfort
zobrazuje výstrahu a vydávánepřetržité pípání.

Viz tabulka „Druhy alarmů“.

Viz tabulka „Druhy alarmů“.

Kyslíkový koncentrátor
se nezapne.

Baterie je vybitá.

Zapojte přístroj do zásuvky se střídavým nebo
stejnosměrným proudem.

Přístroj není připojen
k externímu napájení.
Obecná porucha.

Přesvědčte se, že jsou externí přípojky zajištěné.
Proveďte kompletní cyklus odpojení napájení –vyjměte baterii a odpojte všechny napájecí
zdroje pro střídavý a stejnosměrný proud.
Následně dokončete cyklus opětovnou instalacíbaterie. V případě potřeby znovu připojte externízdroj napájení.
Kontaktujte dodavatele přístroje a podle potřeby
přejděte na jiný zdroj kyslíku.

Všechny ostatní problémy.

Proveďte kompletní cyklus odpojení napájení –vyjměte baterii a odpojte všechny napájecí
zdroje pro střídavý a stejnosměrný proud.
Následně dokončete cyklus opětovnou instalacíbaterie. V případě potřeby znovu připojte externízdroj napájení.
Podle dostupnosti přejděte na jiný zdroj kyslíku
a kontaktujte dodavatele přístroje.

Poznámka: Některé výstrahy způsobí vypnutí přístroje.
Poznámka: Během úplného výpadku napájení dojde k resetu výstražného systému (všechny předchozí výstrahy
se vymažou) a výstraha se zobrazí pouze tehdy, když po obnovení napájení daný stav stále existuje.

470- CZ PN MN232-C4 L | Uživatelská příručka

FreeStyle Comfort
Informace o cestováníPřístroj FreeStyle Comfort splňuje předpisy
Federální letecké správy (FAA) pro použití napalubě komerčních letů. Před každou cestou(letadlem, vlakem, lodí atd.) byste měli:• zajistit, aby byl koncentrátor v dobrém
provozním stavu,• kontaktovat dopravní společnost alespoň
48 hodin před odletem/odjezdem a informovat
ji o faktu, že budete cestovat s přístrojem
FreeStyle Comfort.
Na cestách se doporučuje mít vždy s sebou
náhradní baterie. Při cestování letadlem počítejte
s dostatečným nabitím baterie kvůli časustrávenému na letišti (odbavení před odletem,
bezpečnostní prohlídka atd.) i případným
mezipřistáním. Většina leteckých společnostívyžaduje, abyste měli s sebou dostatečnémnožství plně nabitých baterií, které poskytnou
napájení přístroje na minimálně 150 % maximálníočekávané délky trvání celé cesty (délka letů,letištní záležitosti před odletem a po příletu,
následné spoje a neočekávaná zpoždění).
Poznámka: V případě cestování letadlem
doporučujeme mít s sebou náhradní bateriev zabezpečené přenosné tašce.

1. Příruční kyslíkový koncentrátor zapněte POUZEs baterií, když není připojen žádný externí zdroj
napájení.

2. Aktivujte režim Letadlo s použitím jednoho
z následujících způsobů:a. Postupujte podle pokynů pro aktivování režimu
Letadlo v části Pokyny pro navigaci v menu.

b. Stiskněte a držte tlačítka Nahoru i Dolů po
dobu 5 sekund, dokud neuslyšíte dlouhé pípnutía dokud neuvidíte, že se objevil symbol letadla.

PŘEDPISY FEDERÁLNÍ LETECKÉSPRÁVY (FAA)

Náhradní baterie musí být jednotlivě chráněny proti
zkratům uložením do původního obalu, v němžbyly zakoupeny, nebo jinou izolací jejich pólů,např. obalením odhalených pólů páskou nebovložením každé baterie do samostatné plastové
tašky či ochranného vaku. Baterie se smí převážet
pouze v příručním zavazadle.

3. Když je zobrazen symbol letadla, zapojtepřenosný kyslíkový koncentrátor do sítě.

AKTIVACE REŽIMU LETADLO
Napájení bateriíPříruční kyslíkový koncentrátor lze v letadlepoužívat běžným provozním způsobem POUZEv případě napájení z baterie, kdy není připojenžádný externí zdroj napájení.
Externí napájení
Pokud je v letadle dostupná elektrická zásuvka
umístěná tak, že ji lze použít jako zdroj napájení
příručního kyslíkového koncentrátoru, můžeposloužit jako záložní řešení pro baterie. V tomtopřípadě je nutné dodržovat následující doporučený
postup pro přechod z napájení baterií na napájení
z elektrické sítě letadla:

Poznámka: Kvůli omezením na palubách letadel
nedochází v režimu Letadlo k nabíjení baterií.

PN MN232-C4 L | Uživatelská příručka CZ - 471

FreeStyle ComfortPříslušenstvíVLOŽENÍ PŘÍSTROJE DO PŘENOSNÉTAŠKYZa účelem dosažení správného výkonu a bezpečnostipoužívejte pouze níže uvedené příslušenství
společnosti CAIRE dodávané prostřednictvím vašeho
dodavatele kyslíku. Použití příslušenství, které neníuvedeno níže, by mohlo nepříznivě ovlivnit výkon a/nebo bezpečný provoz koncentrátoru.

1. Vložte přístroj do spodní části tašky a řádně
zarovnejte filtry a otvory baterie.

Standardní příslušenství přístroje FreeStyle
ComfortPřenosná taška

MI396-1

Batoh

MI459-1

Ramenní popruh

MI397-1

Napájecí zdroj pro střídavý proud(včetně všech kabelů)PW036-1S

Napájecí kabel pro stejnosměrný
proud

CD041-1

Externí filtr hrubých částic (pravý)FI226-1

Externí filtr hrubých částic (levý)FI227-1

2. Zajistěte pomocí zipu kolem spodní části vaku.

Dostupné volitelné příslušenstvíZáložní 8článková baterie

BT034-1S

16článková baterie

BT035-1S

Stolní nabíječka bateriíBT036-1Přenosná taška pro veškeré
příslušenstvíMI372-2

Poznámka: Další možnosti mohou být dostupné
pro napájecí kabely v konkrétních zemích (je-li
uvedeno výše). Potřebujete-li objednat alternativnípříslušenství, kontaktujte společnost CAIRE nebo
svého dodavatele kyslíku.

Pokyny k přenosné tašce
Za účelem vložení přístroje FreeStyle Comfort dopřenosné tašky a jeho zajištění dodržujte následující
pokyny.

472- CZ PN MN232-C4 L | Uživatelská příručka

INSTALACE RAMENNÍHO POPRUHU
1. Ramenní popruh lze připevnit buď přímo
k přístroji, nebo k přenosné tašce.
2. Připněte popruh ke kroužku na přístroji nebona tašce.

FreeStyle Comfort
Likvidace a opětovné použití
Zařízení FreeStyle Comfort včetně všech součástí
vždy vraťte dodavateli zařízení k řádné likvidaci.
Můžete také kontaktovat místní městský úřada požádat o pokyny k řádné likvidaci baterie.
Jednotku lze opakovaně používat různými
osobami, ale je třeba ji vrátit dodavateli zařízení
k opětovnému zpracování.

Směrnice OEEZ a RoHS
Tento symbol má majitelům zařízení
připomenout, že jej mají podle směrniceo odpadních elektrických a elektronických
zařízeních (OEEZ) na konci životnosti odevzdat
sběrnému recyklačnímu zařízení.

Materiály, se kterými se uživatel
dostane do přímého kontaktu• Kryt koncentrátoru: PC/ABS.
• Ovládací panel koncentrátoru: polyester EBG7
nebo jeho ekvivalent.
• Výstup kyslíku, tryska: hliník.
• Štítek jednotky: lexan.
• Konektory kabelu: polykarbonát/vinylchlorid.
• Napájecí kabel: PVC, kov.
• Napájecí zdroj: lexan 940 (polykarbonát).
• Baterie: PC/ABS.
• Štítky baterie a napájecích zdrojů:polyesterová fólie.
• Přenosná taška koncentrátoru: cordura, neopren,nylonová podšívka.
• Popruh: polypropylen, acetyl, vinyl, cordura,
neopren.

Naše výrobky odpovídají směrnici o omezenípoužívání některých nebezpečných látek
v elektrických a elektronických zařízeních (RoHS).
Neobsahují více než stopové množství olova anijiných nebezpečných materiálů.
UPOZORNĚNÍ: Za účelem řádné
likvidace požádejte o pokyny dodavatelezařízení nebo místní státní úřad.
UPOZORNĚNÍ: Molekulární síto vždy likvidujte
v souladu se všemi federálními, státními a místnímipředpisy pro kontrolu životního prostředí.

PN MN232-C4 L | Uživatelská příručka CZ - 473

FreeStyle Comfort
Testování elektromagnetické kompatibility (EMC)
Pokyny a prohlášení výrobce – elektromagnetické emisePřístroj FreeStyle Comfort je určen k použití v níže specifikovaném elektromagnetickém prostředí.
Zákazník nebo uživatel přístroje FreeStyle Comfort by měl zajistit, aby byl přístroj používán v takovém prostředí.
Zkouška emisí
RF emiseCISPR 11
RF emiseCISPR 11
Harmonické emise
IEC 61000-3-2
Kolísání napětí /generování kmitavého
signálu IEC 61000-3-3

Kompatibilita
Skupina 1

Elektromagnetické prostředí – pokynyPřístroj FreeStyle Comfort používá radiofrekvenční energii pouze
pro své vnitřní funkce. Jeho vysokofrekvenční emise jsou tedy
velmi nízké a je nepravděpodobné, že by mohly způsobovat
rušení blízkých elektronických zařízení.

Třída BTřída ASplňuje474- CZ PN MN232-C4 L | Uživatelská příručkaPřístroj FreeStyle Comfort je vhodný k použití ve všech zařízeníchvčetně domácností a zařízení přímo napojených na veřejnou síť
nízkého napětí, která napájí budovy užívané za účelem bydlení.

FreeStyle Comfort
Pokyny a prohlášení výrobce – elektromagnetická odolnostPřístroj FreeStyle Comfort je určen k použití v níže specifikovaném elektromagnetickém prostředí. Zákazník nebouživatel přístroje FreeStyle Comfort by měl zajistit, aby byl zásobník používán v takovém prostředí.
Zkouška odolnosti IEC 60601,
Úroveň kompatibility Elektromagnetické prostředí – pokyny
zkušební úroveň
Podlahy by měly být dřevěné, betonové
Elektromagnetickénebo z keramické dlažby. Jsou-li pod±8 kV kontaktní výboj
±8 kV kontaktní výboj
prostředí – pokyny
lahy pokryty syntetickým materiálem,±15 kV vzduch±15 kV vzduch
relativní vlhkost by měla dosahovat
IEC 61000-4-2
alespoň 30 %.
±2 kV pro napájecí
±2 kV pro napájecí
Rychlé elektrické
Kvalita zdroje napájení by měla být
vedení
vedení
přechodné jevy /na úrovni typického profesionálního
skupiny impulzů
zdravotnického prostředí a typického±1 kV pro vstupní /prostředí pro domácí zdravotní péči.
IEC 61000-4-4
Nehodí sevýstupní vedení±1 kV mezi vedeními

Rázový impulz
IEC 61000-4-5±2 kV v běžném režimu
na střídavém vedení±1 kV mezi vedeními±1 kV v diferenčnímrežimu na střídavém
vedení

Kvalita zdroje napájení by měla být
na úrovni typického profesionálního
zdravotnického prostředí a typického
prostředí pro domácí zdravotní péči.

±2 kV v běžném režimu
na vnějším vst./výst.
vedení
0 % UT po dobu 0,5 cyk- 0 % UT po dobu 0,5 cyklu (0°, 45°, 90°, 135°,lu (0°, 45°, 90°, 135°,
180°, 225°, 270°, 315°) 180°, 225°, 270°, 315°) Kvalita zdroje napájení by měla být
na úrovni typického profesionálního
zdravotnického prostředí a typického
Krátkodobé poklesy
0 % UT po dobu 1 cyklu 0 % UT po dobu 1 cyklu prostředí pro domácí zdravotní péči.
napětí, krátká(0°)
přerušení a pomalé (0°)
Pokud uživatel zařízení FreeStylezměny napětí na
Comfort vyžaduje nepřetržitý provozpřívodním vedeníběhem přerušení dodávky elektrické
70 % UT 30% pokles
70 % UT 30% pokles
napájecích zdrojů, v U ) po dobu 25/30 cyk- v U ) po dobu
energie, doporučujeme, aby bylo
TT
IEC 61000-4-11
zařízení FreeStyle Comfort napájeno
lů (0°)
25/30 cyklů (0°)
z nepřerušitelného zdroje napájení nebo
z baterie.
0 % UT po dobu
0 % UT po dobu
250/300 cyklů (0°)
250/300 cyklů (0°)
Magnetická pole síťové frekvence by
Magnetické poleměla být na úrovních charakteristických
síťového kmitočtu
3 A/m
3 A/m
pro typické místo v profesionálním
(50/60 Hz)
zdravotnickém prostředí nebo prostředí
IEC 61000-4-8
pro domácí zdravotní péči.
POZNÁMKA: UTstřídavé napětí sítě před použitím zkušební úrovně.
POZNÁMKA: V určitých případech může dojít k restartování přístroje FreeStyle Comfort v reakci na výrazný
elektrostatický výboj. Za účelem obnovení běžného provozu může být nutné provést kompletní cyklus napájení.
Podrobnější informace najdete v oddílu „Řešení problémů“.

PN MN232-C4 L | Uživatelská příručka CZ - 475

FreeStyle Comfort
Pokyny a prohlášení výrobce – elektromagnetická odolnostPřístroj FreeStyle Comfort je určen k použití v níže specifikovaném elektromagnetickém prostředí. Zákazník nebouživatel zařízení FreeStyle Comfort by měl zajistit, aby byl zásobník používán v takovém prostředí.
Zkouška
IEC 60601,
Úroveň kompatibility
Elektromagnetické prostředí – pokyny
odolnosti
zkušební úroveň
10 Vrms
RF vedení
10 VrmsPřenosná a mobilní radiofrekvenční
komunikační zařízení by se neměla
150 kHz až 80 MHz
IEC 61000-4-6používat v menší vzdálenosti od jakékoli
části přístroje FreeStyle Comfort včetně
kabelů, než je doporučená oddělovací
vzdálenost vypočítaná z rovnice platné
10 V/mRF záření
10 V/mpro frekvenci vysílače.
IEC 61000-4-3
80 MHz až 2700 MHz
Doporučená oddělovací vzdálenostd = 1.2 P
d = 1.2 P = 80 MHz až 800 MHz
d = 1.2 P = 800 MHz až 2700 MHz
kde P je maximální jmenovitý výstupnívýkon vysílače ve wattech (W) udávaný
výrobcem vysílače a d je doporučenáoddělovací vzdálenost v metrech (m).
Intenzita pole pevných radiofrekvenčních
vysílačů zjištěná elektromagnetickým
průzkumem lokalita a by měla býtnižší než povolená úroveň v každém
z frekvenčních pásem. b
K rušení může docházet v blízkostizařízení označených následujícím
symbolem:POZNÁMKA 1: U frekvencí 80 MHz a 800 MHz platí vyšší frekvenční rozsah.
POZNÁMKA 2: Tyto pokyny nemusí platit ve všech situacích. Na šíření elektromagnetických vln má vliv míra jejichpohlcování budovami, předměty a lidmi i míra jejich odrazu od nich.
aIntenzitu pole pevných vysílačů, například základnových stanic radiových (mobilních/bezdrátových) telefonů
a pozemních mobilních radiostanic, amatérských vysílaček, rozhlasového vysílání na AM a FM frekvencích
a televizního vysílání, nelze předem teoreticky přesně stanovit. K posouzení elektromagnetického prostředí
z hlediska přítomnosti pevných vysokofrekvenčních vysílačů je třeba zvážit provedení elektromagnetickéhoprůzkumu lokality. Přesahuje-li naměřená intenzita pole v lokalitě, kde je zařízení FreeStyle Comfort používáno,příslušnou výše uvedenou radiofrekvenční úroveň, je třeba ověřit běžný provoz přístroje FreeStyle Comfort.
Vykazuje-li přístroj FreeStyle Comfort během provozu odchylky, může být nutné provedení určitých opatření,např. změna orientace přístroje nebo jeho přemístění.
b

Ve frekvenčním rozsahu 150 kHz až 80 MHz by měla být intenzita pole nižší než 10 V/m.

476- CZ PN MN232-C4 L | Uživatelská příručka

FreeStyle Comfort
Doporučené oddělovací vzdálenosti mezi přenosnými a mobilními radiofrekvenčními
komunikačními zařízeními a přístrojem FreeStyle ComfortPřístroj FreeStyle Comfort je určen k použití v elektromagnetickém prostředí, v němž je vyzařované radiofrekvenční
rušení regulováno. Zákazník nebo uživatel přístroje FreeStyle Comfort může předcházet elektromagnetickému
rušení tím, že bude dodržovat níže uvedenou minimální vzdálenost mezi přenosnými a mobilními radiofrekvenčními
komunikačními zařízeními (vysílači) a přístrojem FreeStyle Comfort podle maximálního výstupního výkonu
komunikačního zařízení.
Jmenovitý maximální
Vzdálenost podle frekvence vysílače
výstupní výkon
mvysílače
150 kHz až 80 MHz
80 až 800 MHz
800 MHz až 2,5 GHz

d = 1.2 P
d = 1.2 P
d = 1.2 P
W
0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
U vysílačů, jejichž jmenovitý maximální výstupní výkon není uveden výše, lze doporučenou oddělovací vzdálenostd v metrech (m) odhadnout s použitím rovnice příslušné pro frekvenci daného vysílače, kde P je maximální
jmenovitý výstupní výkon vysílače ve wattech (W) udávaný výrobcem vysílače.
POZNÁMKA 1: U frekvencí 80 MHz a 800 MHz platí vzdálenost pro vyšší frekvenční rozsah.
POZNÁMKA 2: Tyto pokyny nemusí platit ve všech situacích. Na šíření elektromagnetických vln má vliv míra jejichpohlcování budovami, předměty a lidmi i míra jejich odrazu od nich.

Klasifikace

Druh ochrany před úrazem elektrickým proudem:Třída II	Ochrany před úrazem elektrickým
proudem je dosaženo dvojitou izolací.
Stupeň ochrany před úrazem elektrickým proudem:Zařízení	typu BF zajišťuje konkrétní		stupeň ochrany před úrazem elektrickým
proudem týkající se		1) přípustného svodového proudu,		2) spolehlivosti ochranného uzemnění
		(je-li k dispozici).
		Není určeno k přímému použití na srdci.
Nezávislé testování podle normy pro zdravotnické
elektrické přístroje:	Testováno za účelem souladu s předpisy• IEC 60601-1, vydání 3.1: Zdravotnické
elektrické přístroje – část 1: Všeobecné požadavky na bezpečnost.
• CAN/CSA C22.2 No. 60601-1-8, vydání 2.1,
Zdravotnické elektrické přístroje – část 1:
Všeobecné požadavky na bezpečnost.

	• Testováno za účelem souladu s předpisem
RTCA/DO160, část 21, kategorie M.
• CISPR 11 / EN 55011, třída B, skupina 1,	„Průmyslová, vědecká a lékařská (ISM) zařízení“• FCC, část 15, oddíl B – třída B Neúmyslné
zdroje záření.
IP 22 – Ochrana před vniknutím prstů nebo
podobných těles s průměrem větším než
12,5 mm. Svisle kapající voda by neměla mítškodlivý vliv, pokud je přístroj nakloněn podúhlem 15° od své běžné polohy.
Povolené metody čištění a kontroly infekcí:Viz část „Čištění, péče a pravidelná údržba“této uživatelské příručky k přístroji FreeStyle.
Stupeň bezpečnosti použití za přítomnostihořlavých anestetických plynů:	Přístroj není určen k tomuto druhu použití.
Provozní režim: Nepřetržitý provoz.

Ochrana před potenciálním elektromagnetickým
nebo jiným rušením mezi přístrojem a jinými
zařízeními.
• Testováno za účelem souladu s normou
EN 60601-1-2 /IEC 60601-1-2: 4. vydání.
PN MN232-C4 L | Uživatelská příručka CZ - 477

FreeStyle Comfort
Elementy sterowania i wskaźniki stanu systemu
ISO 7000Podręcznik operatora; instrukcja
obsługi. Reg. # 1641

ISO 7010: Symbole graficzne — kolory i znaki
ostrzegawcze — zarejestrowane znaki ostrzegawcze
Przeczytać instrukcję obsługi.
Reg. # M002

Kraj i data produkcji. Skrót „CC”
stanowi dwuliterowy kod kraju
produkcji. Data produkcji jest w
formacie RRRR-MM-DD. Reg. # 6049

Przechowywać z dala od nieosłoniętego
płomienia, ognia i źródeł iskrzenia.
Chronić przed źródłem zapłonu.
Zakaz palenia. Reg. # P003

Chronić przed deszczem. Przechowywać
w suchym miejscu. Reg. # 0626

Nie palić w pobliżu urządzenia ani
podczas jego obsługi. Reg. # P002

Maksymalna liczba składanych
jednostek, gdzie „n” wskazuje
maksymalną dozwoloną liczbę
jednostek. Reg. # 2403
Numer katalogowy. Reg. # 2493

Ostrzeżenie. Reg. # W001
IEC 60417

Numer seryjny. Reg. # 2498

Część klasy BF (stopień ochrony przedporażeniem elektrycznym). Reg. # 5333

Zakres temperatury przechowywania lub
temperatury roboczej. Reg. # 0632

Urządzenie klasy II, z podwójną
izolacją. Reg. # 5172

Zakres wilgotności w miejscu
przechowywania. Reg. # 2620

Rozporządzenie UE 2017/745
Autoryzowany przedstawiciel w EuropieJeżeli etykieta z niepowtarzalnym
kodem identyfikacyjnym wyrobu (UDI)
zawiera symbol CE####, oznacza
to, że urządzenie spełnia wymaganiaokreślone dla wyrobów medycznych.
Symbol CE#### wskazuje numer
jednostki notyfikowanej.

Limit ciśnienia atmosferycznego.
Reg. # 2621
Nazwa i adres producenta. Reg. # 3082
Góra. Reg. # 0623

Dodatkowe symbole
Przechowywać z dala od materiałówłatwopalnych, olejów i smarów.

Ostrożnie. Reg. # 0621

Nie rozbierać.

Zawiera substancje niebezpieczne.
Reg. # 3723

Zwiększanie przepływu.

Importer. Reg. # 3725

Zmniejszanie przepływu.

Wylot O2

Przycisk menu.
Przełącznik zasilania. Reg. # 5009
Zgodność
z RTCA/
DO-160,
sekcja 21,
kategoria M

478- PL PN MN232-C4 L | Instrukcja obsługi

Producent niniejszego POC ustalił, że
wyrób ten jest zgodny ze wszystkimi
obowiązującymi kryteriami akceptacji
FAA i użycia na pokładzie statku
powietrznego.

FreeStyle Comfort
Producent niniejszego POC ustalił, że
wyrób ten jest zgodny ze wszystkimi
obowiązującymi kryteriami akceptacji
FAA i użycia na pokładzie statku
powietrznego.

Dyrektywa Rady 2012/19/UE
WEEE
Ten symbol przypomina właścicielomsprzętu, że zgodnie z Dyrektywą w
sprawie zużytego sprzętu elektrycznego
i elektronicznego (WEEE) pozakończeniu eksploatacji należy go oddać
do zakładu utylizacji.
Nasze produkty są zgodne z Dyrektywą
w sprawie ograniczenia stosowania
niektórych niebezpiecznych substancji wsprzęcie elektrycznym i elektronicznym(RoHS). Zawierają jedynie śladoweilości ołowiu i innych niebezpiecznych
materiałów.

Symbol recyklingu
Autoryzowany przedstawiciel
w Szwajcarii (ADD)Jeżeli wyrób jest opatrzony znakiem
UKCA, jak przedstawiono za pomocą
UKCA#### wskazującego numer
jednostki notyfikowanej, to urządzenie
to jest zgodne z przepisami UKCA.
21 CFR 801.15
Prawo federalne dopuszcza sprzedaż
tego urządzenia wyłącznie przez lekarza
lub na jego zlecenie

IEC 60601-1-8
Alarm techniczny o niskim priorytecie

Certyfikat ETL
Dopuszczone do obrotu w Stanach
Zjednoczonych i Kanadzie. Zgodnyz mającymi zastosowanie normami
w Stanach Zjednoczonych i Kanadzie
Zawiera ID FCC: WAP2001
Informacja FCC
To urządzenie może zawierać CYBLE-022001-00,łącznie z anteną 2450AT18B100 firmy Johnson
Technology, spełnia wymagania par. 15 przepisów
FCC. Urządzenie spełnia wymagania dotyczące
homologacji nadajnika modułowego zgodnie
z informacją publiczną FCC DA00-1407. Działanie
nadajnika podlega następującym dwóm warunkom(1): urządzenie nie może powodować szkodliwych
zakłóceń oraz (2) urządzenie musi przyjmować
wszelkie odbierane zakłócenia, w tym zakłócenia,
które mogą powodować niepożądane działanie.
Produkt może być chroniony jednym lub wieloma
patentami w Stanach Zjednoczonych i innych
krajach. Odpowiednie patenty wyszczególnione
są w naszej witrynie pod adresem podanym poniżej.
Patenty: www.caireinc.com/corporate/patents
Symbole graficzne stosowane w urządzeniach wg
IEC 60417

ISO 15223-1
Wyrób medyczny. Reg. # 5.7.7
Kod niepowtarzalnego identyfikatora
wyrobu 5.7.10
A S/NZS 4417.1, AS/NZS 4417.2
Znak zgodności z przepisami ACMA —wskazuje zgodność z przepisami ACMA
(ADD)

Technologia bezprzewodowa
Rodzaje połączeń

SSP, iAP2, GATTCzęstotliwość

od 2402 do 2480 MHz

Maks. moc wyjściowa RF+4 dBmZasięg działania

10 m (klasa 2)

Zaleca się, aby podczas pracy urządzenie znajdowało
się przynajmniej 1,1 cm (0,43 in.) od ciała. Nie dotyczy
masek, przewodu powietrza ani akcesoriów.

Prąd stały. Reg. # 5031
Prąd przemienny. Reg. # 5032
IEC 60601-1
Sprzęt kroploszczelny

PN MN232-C4 L | Instrukcja obsługi PL - 479

FreeStyle Comfort
Rozpakowywanie koncentratora FreeStyle Comfort
Upewnić się, że w opakowaniu
znajdują się wszystkie
wyszczególnione komponenty.
Jeśli stwierdzone zostaną braki,należy natychmiast skontaktowaćsię z dostawcą rozwiązania do
podawania tlenu.

Akcesoria standardowe• Przenośny koncentrator tlenu• Torba• Pasek na ramię
• Akumulator 8-ogniwowy• Zasilacz prądu przemiennego(AC)• Przewód zasilania AC• Zasilacz DC• Instrukcja obsługi

Akcesoria opcjonalne• Dodatkowy akumulator
8-ogniwowy• Akumulator 16-ogniwowy• Ładowarka biurkowa• Torba na wszystkie akcesoria

Podstawowe informacje na temat koncentratora FreeStyle ComfortPoniżej opisano ważne części oraz interfejs użytkownika koncentratora FreeStyle Comfort.

Widok z przodu
Ucho mocowania paska

InterfejsużytkownikaZłączewylotowe
tlenu
Wylot
powietrza

Wlot/filtry
powietrza
Wlot/filtry
powietrza

Zatyczka
gniazdazasilania
Akumulator
Przycisk zwalniania
akumulatora
480- PL PN MN232-C4 L | Instrukcja obsługi

FreeStyle Comfort
Widok z góryWyświetlacz
Zwiększanie
przepływu

Przycisk zasilania

Przycisk menu

Zmniejszanie
przepływu

Wyświetlacz na akumulatorze
Ustawienie przepływuWskaźnik stanu
akumulatoraWskaźnik autoSAT
Procentowy
poziomnaładowania
akumulatoraWyświetlacz
alarmuWskaźnikdostarczania
bolusaWyświetlacz na zasilaczu AC/DC
Ustawienie
przepływu

Miejscewyświetlania
alarmuWskaźnik stanu
akumulatoraWskaźnikładowania

PN MN232-C4 L | Instrukcja obsługi PL - 481

FreeStyle Comfort
Czym jest koncentrator tlenu?
Powietrze, którym oddychamy, zawiera tlen
(około 21%), azot (78%) i inne gazy (1%). Urządzenie
FreeStyle® Comfort® za pomocą specjalnych wlotów
pobiera powietrze z otoczenia. Następnie jest ono
filtrowane przez materiał absorbujący zwany sitemcząsteczkowym. Zatrzymuje on azot, przepuszczając
jedynie tlen. W efekcie użytkownikowi podawane jest
powietrze z wysokim stężeniem tlenu.
FreeStyle Comfort to połączenie koncentratora ioszczędzacza tlenu, którego efektem jest przenośne
urządzenie zapewniające maksymalną mobilność
i aktywność. Zaawansowany system podawania
impulsowego szybko wykrywa, kiedy użytkownik
wdycha powietrze, i podaje impuls tlenu na początku
każdego wdechu. Podawanie impulsowe doskonale
sprawdza się, gdy użytkownik jest aktywny lub
znajduje się poza domem. W odróżnieniu od trybu
ciągłego przepływu pozwala oszczędzać tlen i wydłuża
czas pracy na akumulatorze. Urządzenie FreeStyle
Comfort jest na tyle lekkie, że można je nosić na pasku
na ramieniu. Może być zasilane prądem AC (gniazdościenne), DC (samochód) lub z akumulatora.

Przeznaczenie
Koncentrator tlenu CAIRE FreeStyle Comfort jest
przeznaczony do podawania tlenu. Urządzenie nie jest
przeznaczone do podtrzymywania życia. Nie zapewniateż żadnych narzędzi do monitorowania pacjenta.

Dlaczego lekarz przepisał
podawanie tlenu?
Wiele osób cierpi na różne choroby serca, płuc i dróg
oddechowych. Niejednokrotnie tlenoterapia może im
pomóc w codziennym życiu w domu i w podróży.
Tlen to gaz stanowiący nawet 21% powietrza, którym
oddychamy. Jego ciągłe dostarczanie jest warunkiem
prawidłowego funkcjonowania naszego organizmu.
Lekarz przepisuje ustawienie przepływu tlenu
dostosowane do choroby układu oddechowego, naktórą cierpi użytkownik.

482- PL PN MN232-C4 L | Instrukcja obsługi

Pomimo że tlen jest lekiem nieuzależniającym,samowolne stosowanie go może być niebezpieczne.
Przed użyciem koncentratora tlenu należy zasięgnąć
porady lekarza. Dostawca sprzętu do podawania tlenu
zaprezentuje, jak ustawić przepisany przepływ gazu.

Profil operatora
Koncentratory służą do suplementacji tlenu uużytkowników doświadczających dyskomfortu z
powodu schorzeń wpływających na zdolność płuc
do transportowania tlenu z powietrza do układu
krwionośnego. W przenośnych koncentratorach tlenu
nie przechowuje się tlenu. Nie trzeba ich uzupełniać,a ładować je można wszędzie tam, gdzie dostępna
jest sieć AC lub DC (z wyjątkiem samolotów).
Dzięki temu użytkownicy mogą w dużym stopniu
samodzielnie obsługiwać urządzenie oraz prowadzić
bardziej aktywny i mobilny tryb życia w domu i pozanim. Z koncentratora tlenu można korzystać wyłącznie
na zlecenie lekarza.
Użytkownikami koncentratorów tlenu są zazwyczaj
osoby w wieku powyżej 65 lat cierpiące na różne
choroby układu oddechowego, na przykład przewlekłą
obturacyjną chorobę płuc (POChP). Użytkownik
powinien mieć dobre zdolności poznawcze
i musi być w stanie informować o potencjalnieodczuwanym dyskomforcie. Jeśli użytkownik nie
jest zdolny do informowania o dyskomforcie lub
nie jest w stanie przeczytać ani zrozumieć etykiet
i instrukcji koncentratora, zaleca się, aby użytkowanie
odbywało się pod nadzorem osoby posiadającej takie
zdolności. Jeśli podczas korzystania z koncentratora
odczuwany jest dyskomfort, należy skontaktowaćsię z pracownikiem służby zdrowia. Zaleca się, aby
na wypadek braku zasilania lub awarii koncentratoraużytkownik wyposażył się w zapasowe źródło tlenu(np. tlen w butli). Do korzystania z koncentratora niesą potrzebne żadne inne specjalne umiejętności.

FreeStyle ComfortWażne!
Definicje instrukcji bezpieczeństwa:
OSTRZEŻENIE: WAŻNA INFORMACJA
DOTYCZĄCA ZAGROŻEŃ, KTÓRE MOGĄSPOWODOWAĆ POWAŻNE OBRAŻENIA.
PRZESTROGA: Ważna informacja
pozwalająca uniknąć uszkodzeniaurządzenia FreeStyle Comfort.
Uwaga: Informacja wymagająca szczególnej uwagi.

Wskazania do zastosowania
OSTRZEŻENIE: ZAWSZE USTAWIAĆ
PRZEPISANY PRZEPŁYW TLENU.
USTAWIENIE PRZEPŁYWU MOŻNAZMIENIAĆ TYLKO NA POLECENIE LEKARZA.
OSTRZEŻENIE: URZĄDZENIA NIE TESTOWANOANI NIE ZATWIERDZONO DO STOSOWANIAU PACJENTÓW PEDIATRYCZNYCH.
OSTRZEŻENIE: PRAWO FEDERALNE (STANYZJEDNOCZONE) DOPUSZCZA SPRZEDAŻ TEGOURZĄDZENIA WYŁĄCZNIE PRZEZ LEKARZA LUB
PRACOWNIKA SŁUŻBY ZDROWIA ALBO NA JEGOZLECENIE.
OSTRZEŻENIE: URZĄDZENIE TO NIE JEST
SYSTEMEM PODTRZYMYWANIA ŻYCIA.
UŻYTKOWNICY, KTÓRZY NIE SĄ W STANIEPOWIADAMIAĆ O ODCZUWANYM DYSKOMFORCIEW CZASIE KORZYSTANIA Z KONCENTRATORATLENU, MOGĄ WYMAGAĆ DODATKOWEGOMONITOROWANIA. UŻYTKOWNICY Z WADAMISŁUCHU LUB WZROKU MOGĄ WYMAGAĆ POMOCYW MONITOROWANIU ALARMÓW.
OSTRZEŻENIE: USTAWIENIA TRYBU IMPULSOWEGONALEŻY SKONFIGUROWAĆ DLA KAŻDEGO
UŻYTKOWNIKA ODDZIELNIE, BIORĄC POD UWAGĘJEGO POTRZEBY PODCZAS ODPOCZYNKU,ĆWICZEŃ I W PODRÓŻY.
OSTRZEŻENIE: NINIEJSZE URZĄDZENIE JEST
PRZEZNACZONE DO STOSOWANIA W WARUNKACHOPIEKI DOMOWEJ I INNYCH TEGO TYPUŚRODOWISKACH POZASZPITALNYCH.

OSTRZEŻENIE: CZASAMI CZYNNOŚĆ ODDECHOWA
MOŻE NIE WYWOŁAĆ AKTYWACJI FUNKCJI
OSZCZĘDZANIA.
OSTRZEŻENIE: PRODUKTU NIE NALEŻY STOSOWAĆU PACJENTÓW PO TRACHEOTOMII.
OSTRZEŻENIE: KOBIETY W CIĄŻY ANI KARMIĄCE NIE
POWINNY STOSOWAĆ AKCESORIÓW ZALECANYCHW TEJ INSTRUKCJI, PONIEWAŻ MOGĄ ONEZAWIERAĆ FTALANY.

Przeciwwskazania do zastosowania
OSTRZEŻENIE: W NIEKTÓRYCHOKOLICZNOŚCIACH STOSOWANIE TLENU
BEZ ZALECENIA LEKARZA MOŻE BYĆ
NIEBEZPIECZNE. UŻYWAĆ URZĄDZENIA,
TYLKO JEŚLI ZOSTAŁO PRZEPISANEPRZEZ LEKARZA.
OSTRZEŻENIE: NINIEJSZE URZĄDZENIE NIE JEST
PRZEZNACZONE DO UŻYTKU W STREFIE BADAŃ
METODĄ REZONANSU MAGNETYCZNEGO, TAKIMJAK SYSTEM MRI, ANI W JEGO POBLIŻU.
OSTRZEŻENIE: NIE STOSOWAĆ W OBECNOŚCI
ŁATWOPALNYCH ŚRODKÓW ZNIECZULAJĄCYCH.
OSTRZEŻENIE: KONCENTRATOR JEST
URZĄDZENIEM ELEKTRYCZNYM, W ZWIĄZKUZ CZYM NIE DZIAŁA PODCZAS PRZERW W
ZASILANIU. MOŻE TEŻ WYMAGAĆ SERWISOWANIAPRZEZ WYKWALIFIKOWANEGO SPECJALISTĘ.
NIE STOSOWAĆ KONCENTRATORA TLENU UUŻYTKOWNIKÓW, W PRZYPADKU KTÓRYCH TEGORODZAJU PRZERWY W DZIAŁANIU MOGŁYBY
DOPROWADZIĆ DO NIEPOŻĄDANYCH SKUTKÓWZDROWOTNYCH.
OSTRZEŻENIE: NINIEJSZE URZĄDZENIE NIE JESTZATWIERDZONE DO STOSOWANIA W KAŻDYM
ŚRODOWISKU. EKSPOZYCJA NA NIEKTÓRE
ŹRÓDŁA FAL ELEKTROMAGNETYCZNYCH MOŻESKUTKOWAĆ POWSTAWANIEM ZAKŁÓCEŃ LUB
NIEZAMIERZONEGO DZIAŁANIA URZĄDZENIA.
W PRZYPADKU ZAKŁÓCEŃ URZĄDZENIE NALEŻY
NATYCHMIAST WYŁĄCZYĆ.

OSTRZEŻENIE: TO URZĄDZENIE JEST
PRZEZNACZONE DO UŻYTKU PRZEZ PACJENTA
PODDAWANEGO TLENOTERAPII.

PN MN232-C4 L | Instrukcja obsługi PL - 483

FreeStyle Comfort
Wytyczne dotyczące bezpieczeństwa
OSTRZEŻENIE: MODYFIKOWANIE TEGOURZĄDZENIA JEST ZABRONIONE.
OSTRZEŻENIE: TO URZĄDZENIE DOSTARCZAMIESZANKĘ O WYSOKIM STĘŻENIU TLENU,
KTÓRA POWODUJE SZYBKIE SPALANIE. W
POMIESZCZENIACH, W KTÓRYCH JEST (1) TOURZĄDZENIE LUB ZNAJDUJĄ SIĘ (2) AKCESORIAZAWIERAJĄCE TLEN, NIE WOLNO PALIĆ ANIUŻYWAĆ OTWARTEGO OGNIA. ZIGNOROWANIETEGO OSTRZEŻENIA MOŻE DOPROWADZIĆ DOPOŻARU, USZKODZENIA MIENIA, OBRAŻEŃ CIAŁA,A NAWET ŚMIERCI.
OSTRZEŻENIE: NIE UŻYTKOWAĆ URZĄDZENIA W
OGRANICZONEJ LUB ZAMKNIĘTEJ PRZESTRZENI(NP. MAŁEJ WALIZCE LUB TOREBCE), GDZIEWENTYLACJA JEST UTRUDNIONA. MOŻETO SPOWODOWAĆ PRZEGRZEWANIE SIĘ
KONCENTRATORA TLENU I MIEĆ NEGATYWNYWPŁYW NA JEGO PRACĘ.
OSTRZEŻENIE: USTAWIAJĄC KONCENTRATOR,UNIKAĆ DYMU, ZANIECZYSZCZEŃ POWIETRZAI OPARÓW.
OSTRZEŻENIE: W PRZYPADKU WYSTĄPIENIAPOWAŻNEGO INCYDENTU ZWIĄZANEGO Z
TYM URZĄDZENIEM, UŻYTKOWNIK POWINIENNATYCHMIAST ZGŁOSIĆ GO DOSTAWCY I/
LUB PRODUCENTOWI. POWAŻNY INCYDENTJEST DEFINIOWANY JAKO URAZ, ŚMIERĆ LUBMOŻLIWOŚĆ SPOWODOWANIA OBRAŻEŃ / ŚMIERCIW PRZYPADKU PONOWNEGO WYSTĄPIENIAINCYDENTU. UŻYTKOWNIK MOŻE RÓWNIEŻ ZGŁOSIĆ
INCYDENT WŁAŚCIWEMU ORGANOWI W KRAJU,
W KTÓRYM ZDARZENIE MIAŁO MIEJSCE.
OSTRZEŻENIE: KORZYSTANIE Z
NIEKTÓRYCH AKCESORIÓW DO
PODAWANIA TLENU, KTÓRYCH
NIE WYSZCZEGÓLNIONO WŚRÓD
KOMPONENTÓW DO STOSOWANIA Z TYM
URZĄDZENIEM, MOŻE MIEĆ NEGATYWNYWPŁYW NA JEGO DZIAŁANIE.
ZALECANE AKCESORIA WYMIENIONESĄ W NINIEJSZEJ INSTRUKCJI.
OSTRZEŻENIE: JEŚLI DOSZŁO DO UPUSZCZENIA,
USZKODZENIA LUB ZAMOCZENIA KONCENTRATORATLENU, SKONTAKTOWAĆ SIĘ Z DOSTAWCĄ
SPRZĘTU W CELU SPRAWDZENIA URZĄDZENIAI EWENTUALNEGO DOKONANIA NAPRAWY.
NIE UŻYWAĆ KONCENTRATORA TLENU, JEŚLI
WIDOCZNE JEST USZKODZENIE PRZEWODU LUB
WTYCZKI ZASILANIA.
OSTRZEŻENIE: ABY UNIKNĄĆ USZKODZENIA LUB
OBRAŻEŃ PODCZAS TRANSPORTU W POJEŹDZIE,ODPOWIEDNIO ZABEZPIECZYĆ, PRZYPIĄĆ PASEM LUB
ZAPEWNIĆ INNĄ OCHRONĘ KONCENTRATORA TLENU.
484- PL PN MN232-C4 L | Instrukcja obsługi

OSTRZEŻENIE: PRZEPISY DEPARTAMENTU
TRANSPORTU STANÓW ZJEDNOCZONYCH ORAZ
ORGANIZACJI NARODÓW ZJEDNOCZONYCH(ONZ) WYMAGAJĄ ODŁĄCZENIA AKUMULATORAOD KONCENTRATORA TLENU W PRZYPADKUODPRAWIANIA GO JAKO BAGAŻ PRZED WSZELKIMILOTAMI MIĘDZYNARODOWYMI. PRZYGOTOWUJĄC
KONCENTRATOR DO WYSYŁKI, RÓWNIEŻ NALEŻY
ODŁĄCZYĆ I ODPOWIEDNIO ZAPAKOWAĆAKUMULATOR.
OSTRZEŻENIE: PRZED ROZPOCZĘCIEM OBSŁUGI
URZĄDZENIA BEZWZGLĘDNIE PRZECZYTAĆI ZROZUMIEĆ INSTRUKCJĘ OBSŁUGI. JEŚLI
UŻYTKOWNIK NIE JEST W STANIE ZROZUMIEĆ
OSTRZEŻEŃ I INSTRUKCJI, PRZED ROZPOCZĘCIEM
UŻYTKOWANIA SPRZĘTU NALEŻY SKONTAKTOWAĆ
SIĘ Z DOSTAWCĄ. W PRZECIWNYM RAZIE MOŻEDOJŚĆ DO OBRAŻEŃ CIAŁA LUB USZKODZENIAURZĄDZENIA.
OSTRZEŻENIE: W RAZIE ODCZUWANIA
DYSKOMFORTU LUB WYSTĄPIENIA ZAGROŻENIAZDROWIA NATYCHMIAST SKORZYSTAĆ Z POMOCYLEKARSKIEJ.
OSTRZEŻENIE: JEŚLI KONCENTRATORTLENU BĘDZIE UŻYWANY W WARUNKACHWYKRACZAJĄCYCH POZA ZAKRES TEMPERATURROBOCZYCH, GENEROWANE STĘŻENIE TLENU
MOŻE BYĆ NIŻSZE NIŻ W DANYCH TECHNICZNYCH.
INFORMACJE NA TEMAT LIMITÓW TEMPERATUR
DOSTĘPNE SĄ W TEJ INSTRUKCJI, W CZĘŚCIZAWIERAJĄCEJ DANE TECHNICZNE.
OSTRZEŻENIE: DO KONSERWACJI AKCESORIÓWZAWIERAJĄCYCH TLEN ANI DO KONCENTRATORATLENU NIE UŻYWAĆ OLEJU, SMARU, PRODUKTÓW
OPARTYCH NA ROPIE NAFTOWEJ ANI INNYCH
ŚRODKÓW ŁATWOPALNYCH. WYKORZYSTYWAĆ
TYLKO BALSAMY I MAŚCI NA BAZIE WODY, KTÓRE
MOŻNA STOSOWAĆ W OBECNOŚCI TLENU.
TLEN PRZYŚPIESZA SPALANIE ŁATWOPALNYCHSUBSTANCJI.
OSTRZEŻENIE: NIEPRAWIDŁOWE UŻYWANIEAKUMULATORA MOŻE DOPROWADZIĆ DO JEGOROZGRZANIA LUB ZAPŁONU, A W EFEKCIE DOPOWAŻNYCH OBRAŻEŃ. AKUMULATORA NIE WOLNOPRZEBIJAĆ, UDERZAĆ, DEPTAĆ, UPUSZCZAĆ ANI
W INNY SPOSÓB NARAŻAĆ NA SILNE UDARY LUB
WSTRZĄSY. KORZYSTANIE Z USZKODZONEGOAKUMULATORA MOŻE DOPROWADZIĆ
DO OBRAŻEŃ CIAŁA.
OSTRZEŻENIE: NIE ZWIERAĆ METALOWYCH STYKÓWAKUMULATORA METALOWYMI PRZEDMIOTAMI,
TAKIMI JAK KLUCZE CZY MONETY. MOŻE TO
WYGENEROWAĆ ISKRY LUB DUŻĄ ILOŚĆ CIEPŁA.

FreeStyle Comfort
OSTRZEŻENIE: JEŚLI PRZENOŚNY KONCENTRATORTLENU ZNAJDUJE SIĘ NA ZEWNĄTRZ I JEST
ZASILANY PRĄDEM PRZEMIENNYM, PODŁĄCZAĆZASILACZ WYŁĄCZNIE DO GNIAZDA ZWYŁĄCZNIKIEM RÓŻNICOWOPRĄDOWYM.
OSTRZEŻENIE: NIE USTAWIAĆ KONCENTRATORATLENU W POBLIŻU INNEGO SPRZĘTU ANI NA NIM.
JEŚLI NIE MA INNEJ MOŻLIWOŚCI, MONITOROWAĆ
PRAWIDŁOWE DZIAŁANIE URZĄDZENIA.
OSTRZEŻENIE: KORZYSTANIE Z PRZEWODÓW I
ADAPTERÓW INNYCH NIŻ WSKAZANE, Z WYJĄTKIEMPRZEWODÓW I ADAPTERÓW SPRZEDAWANYCHPRZEZ PRODUCENTA MEDYCZNEGO URZĄDZENIAELEKTRYCZNEGO, JAKO CZĘŚCI ZAMIENNYCHDLA KOMPONENTÓW WEWNĘTRZNYCH MOŻESPOWODOWAĆ ZWIĘKSZENIE EMISJI LUB
OBNIŻENIE ODPORNOŚCI KONCENTRATORA TLENUNA ZAKŁÓCENIA.
OSTRZEŻENIE: PODŁĄCZAĆ URZĄDZENIE TYLKO DO
ZASILANIA O NAPIĘCIU WSKAZANYM NA ETYKIECIEZ DANYMI TECHNICZNYMI ZNAJDUJĄCEJ SIĘ NA
KONCENTRATORZE.
OSTRZEŻENIE: NIE PODŁĄCZAĆ URZĄDZENIA ZAPOMOCĄ PRZEDŁUŻACZY. NIE PODŁĄCZAĆ ZBYTWIELU URZĄDZEŃ DO TEGO SAMEGO GNIAZDAZASILANIA. ZASTOSOWANIE PRZEDŁUŻACZY
MOŻE MIEĆ NEGATYWNY WPŁYW NA DZIAŁANIE
KONCENTRATORA. PODŁĄCZENIE ZBYT WIELUURZĄDZEŃ DO JEDNEGO GNIAZDA MOŻESPOWODOWAĆ PRZECIĄŻENIE SKRZYNKI
ROZDZIELCZEJ I W EFEKCIE AKTYWACJĘ
WYŁĄCZNIKA/BEZPIECZNIKA LUB POŻAR,JEŚLI WYŁĄCZNIK/BEZPIECZNIK NIE ZADZIAŁA.
PRZESTROGA: W przypadku alarmu
lub zaobserwowania nieprawidłowego
działania koncentratora tlenu zapoznać
się z częścią tej instrukcji poświęconą
rozwiązywaniu problemów. Jeśli nie uda się
rozwiązać problemu, należy skontaktowaćsię z dostawcą sprzętu.
PRZESTROGA: Przestrzegać instrukcji producenta.
W przeciwnym razie gwarancja będzie nieważna.

OSTRZEŻENIE: NIE PODEJMOWAĆ
PRÓB SERWISOWANIA URZĄDZENIA,Z WYJĄTKIEM POTENCJALNYCHROZWIĄZAŃ PROBLEMÓWWYMIENIONYCH W TEJ INSTRUKCJI.
OSTRZEŻENIE: UŻYWAĆ TYLKO AKCESORIÓWZALECANYCH PRZEZ PRODUCENTA. STOSOWANIEINNYCH MOŻE BYĆ NIEBEZPIECZNE, SPOWODOWAĆ
POWAŻNE USZKODZENIE KONCENTRATORA TLENUI DOPROWADZIĆ DO UNIEWAŻNIENIA GWARANCJI.
OSTRZEŻENIE: PRODUCENT ZALECA, ABY W RAZIEPRZERWY W ZASILANIU, SYTUACJI ALARMOWEJ
LUB AWARII MECHANICZNEJ DOSTĘPNE BYŁOALTERNATYWNE ŹRÓDŁO TLENU. ABY WYBRAĆRODZAJ SYSTEMU REZERWOWEGO, NALEŻYSKONSULTOWAĆ SIĘ Z DOSTAWCĄ SPRZĘTU.
OSTRZEŻENIE: UWAŻAĆ, ABY NIE ZABLOKOWAĆ
WLOTÓW ANI WYLOTU POWIETRZA. NIE WRZUCAĆ
ANI NIE WKŁADAĆ PRZEDMIOTÓW DO OTWORÓW W
OBUDOWIE URZĄDZENIA. MOŻE TO SPOWODOWAĆ
PRZEGRZEWANIE SIĘ KONCENTRATORA TLENU
I MIEĆ NEGATYWNY WPŁYW NA JEGO PRACĘ.
OSTRZEŻENIE: PALENIE PODCZAS UŻYWANIA
TLENU JEST GŁÓWNĄ PRZYCZYNĄ OBRAŻEŃ
I ŚMIERCI SPOWODOWANYCH PRZEZ OGIEŃ.
NALEŻY BEZWZGLĘDNIE PRZESTRZEGAĆ
PONIŻSZYCH OSTRZEŻEŃ BEZPIECZEŃSTWA:
OSTRZEŻENIE: W POMIESZCZENIACH, W KTÓRYCH
JEST TO URZĄDZENIE LUB ZNAJDUJĄ SIĘ
AKCESORIA ZAWIERAJĄCE TLEN, NIE WOLNOPALIĆ, UŻYWAĆ ŚWIECZEK ANI OTWARTEGO OGNIA.
OSTRZEŻENIE: PALENIE PODCZAS STOSOWANIA
WĄSÓW TLENOWYCH MOŻE DOPROWADZIĆ DO
OPARZEŃ TWARZY, A NAWET ŚMIERCI.
OSTRZEŻENIE: PALĄCY UŻYTKOWNICY POWINNIPOSTĘPOWAĆ ZGODNIE Z NASTĘPUJĄCĄPROCEDURĄ, KTÓRA MOŻE IM URATOWAĆ ŻYCIE:WYŁĄCZYĆ KONCENTRATOR TLENU, WYJĄĆ WĄSYI OPUŚCIĆ POMIESZCZENIE, W KTÓRYM ZNAJDUJESIĘ URZĄDZENIE.

PN MN232-C4 L | Instrukcja obsługi PL - 485

FreeStyle Comfort
OSTRZEŻENIE: W DOMU LUB W MIEJSCUPODAWANIA TLENU UMIEŚCIĆ W WIDOCZNYCHMIEJSCACH OSTRZEŻENIE „ZAKAZ PALENIA —PODAWANIE TLENU”. POINFORMOWAĆ
UŻYTKOWNIKÓW I ICH OPIEKUNÓW O
ZAGROŻENIACH ZWIĄZANYCH Z PALENIEMW OBECNOŚCI UŻYTKOWNIKA LUB PODCZASPODAWANIA TLENU MEDYCZNEGO.
OSTRZEŻENIE: NIE UŻYWAĆ KONCENTRATORATLENU W OBECNOŚCI ŁATWOPALNYCH GAZÓW.
MOŻE TO SPOWODOWAĆ NAGŁY ZAPŁON,KTÓREGO EFEKTEM BĘDZIE USZKODZENIE MIENIA,
OBRAŻENIE CIAŁA LUB ŚMIERĆ.
OSTRZEŻENIE: JEŚLI WĄSY ZOSTANĄ WYJĘTEZ NOSA I UMIESZCZONE NA UBRANIU, POŚCIELI,SOFIE LUB INNYM MIĘKKIM MATERIALE, A W
POBLIŻU BĘDZIE ZNAJDOWAĆ SIĘ PAPIEROS,
ŹRÓDŁO CIEPŁA LUB PŁOMIEŃ, MOŻE DOJŚĆ
DO WYBUCHU POŻARU.
OSTRZEŻENIE: NIE ZOSTAWIAĆ WĄSÓW
TLENOWYCH NA UBRANIU, POŚCIELI CZYPODUSZKACH ANI POD NIMI. GDY WŁĄCZONEURZĄDZENIE NIE JEST UŻYWANE, OBECNOŚĆ
TLENU ZWIĘKSZA RYZYKO ZAPALENIA SIĘ
MATERIAŁU. WYŁĄCZAĆ ZASILANIE ZAWSZE,GDY KONCENTRATOR TLENU NIE JEST UŻYWANY.
OSTRZEŻENIE: ZATWIERDZONE AKCESORIA DO
PODAWANIA TLENU MAJĄ NA CELU OGRANICZENIEROZPRZESTRZENIANIA SIĘ OGNIA ZE WZGLĘDU
NA BEZPIECZEŃSTWO PACJENTA. STOSOWANIENIEZATWIERDZONYCH AKCESORIÓW MOŻEZWIĘKSZYĆ RYZYKO OBRAŻEŃ LUB USZKODZEŃ
ZWIĄZANYCH Z POŻAREM.
OSTRZEŻENIE: PRZED ODŁĄCZENIEM
AKUMULATORA NALEŻY WYŁĄCZYĆ
KONCENTRATOR TLENU I ODŁĄCZYĆGO OD ZEWNĘTRZNEGO ŹRÓDŁA ZASILANIA.
OSTRZEŻENIE: MAŁE CZĘŚCI
MOGĄ STWARZAĆ ZAGROŻENIE
ZADŁAWIENIEM.
OSTRZEŻENIE: NIE DOTYKAĆ WYLOTU POWIETRZAPRZEZ DŁUŻEJ NIŻ 10 SEKUND, PONIEWAŻ MOŻEMIEĆ WYSOKĄ TEMPERATURĘ.
OSTRZEŻENIE: WIATR ORAZ SILNE PRZECIĄGI
MOGĄ MIEĆ NEGATYWNY WPŁYW NA PRECYZJĘ
DZIAŁANIA URZĄDZENIA PODCZAS TLENOTERAPII.
OSTRZEŻENIE: STOSOWANIE TEGO PRODUKTUMOŻE NARAŻAĆ NA KONTAKT Z NIKLEM, KTÓRYW STANIE KALIFORNIA UZNAWANY JEST ZA
SUBSTANCJĘ RAKOTWÓRCZĄ. DODATKOWE
INFORMACJE DOSTĘPNE SĄ NA STRONIE
WWW.P65WARNINGS.CA.GOV.

486- PL PN MN232-C4 L | Instrukcja obsługi

PRZESTROGA: Zasilacz AC każdorazowoodłączyć najpierw od gniazda sieciowego,
a dopiero później od koncentratora tlenu.
PRZESTROGA: Koncentrator tlenu ustawić w taki
sposób, aby zachować swobodny dostęp do przewodu
zasilania.
PRZESTROGA: Zadbać o takie rozmieszczenie
przewodów doprowadzania tlenu i zasilania, aby nie
doszło do przypadkowego potknięcia ani uduszenia.
PRZESTROGA: Jeśli koncentrator tlenu używany jest
w samochodzie, na łodzi lub jeżeli urządzenie jestpodłączone do dowolnego źródła zasilania prądem
stałym, pojazd uruchomić przed podłączeniemurządzenia. W przeciwnym razie zasilacz może nie
doprowadzać prądu do koncentratora tlenu.
PRZESTROGA: Po wyłączeniu silnika samochodu,
w którym używany jest koncentrator tlenu, odłączyć
urządzenie i wyjąć je z pojazdu. Nie pozostawiać
koncentratora tlenu w bardzo rozgrzanym aniwyziębionym samochodzie, ani w innych miejscach,w których panuje wysoka lub niska temperatura.
Uwaga: Przenośny sprzęt komunikacyjny wykorzystujący
częstotliwości radiowe może oddziaływać na elektrycznysprzęt medyczny.
PRZESTROGA: NIE pozostawiaćpodłączonego koncentratora tlenu
ani zasilacza, gdy silnik pojazdu jestWYŁĄCZONY. Może to spowodować
wyczerpanie akumulatora pojazdu.
PRZESTROGA: Korzystanie z koncentratora tlenu lub
przechowywanie go w warunkach wykraczających
poza normalny zakres temperatur i ciśnień roboczych(wysokość n.p.m.) może mieć wpływ na jego działanie,
skrócić czas pracy na akumulatorze, a także wydłużyć
czas ładowania akumulatora. Informacje na temat
limitów temperatur roboczych i przechowywaniadostępne są w tej instrukcji, w części zawierającej
dane techniczne.
PRZESTROGA: Koncentratory tlenu wyposażone
w akumulatory należy przechowywać w
chłodnym i suchym miejscu. Pozwoli to wydłużyć
czas eksploatacji akumulatora. Długotrwałe
przechowywanie koncentratora tlenu w miejscu, w
którym panuje wysoka temperatura, lub z całkowicie
naładowanym bądź rozładowanym akumulatoremmoże doprowadzić do skrócenia okresu eksploatacji
ogniw. Nie podejmować prób otwarcia akumulatora.
Nie ma w nim części serwisowanych. Trzymać
akumulatory z dala od dzieci.
PRZESTROGA: UŻYWAĆ WYŁĄCZNIE akumulatorów
dostarczonych przez producenta. Aby dowiedzieć
się, jak prawidłowo zutylizować akumulator, należy
skontaktować się z dostawcą sprzętu lub odpowiednim
organem władz lokalnych.

FreeStyle Comfort
Uwaga: W celu zminimalizowania szkód spowodowanych
przez koncentrator lub awarię zasilania dostawca sprzętu
powinien ocenić zapotrzebowanie na zapasowy tlen
podczas instalacji i okresowo, gdy zmianie ulegają:- stan pacjenta- środowisko, w którym żyje pacjent- możliwość zaopatrzenia pacjenta w zapasowy tlen.
Uwaga: Zadbać, aby wąsy były prawidłowo
zamontowane. Dzięki temu koncentrator tlenu będzie
mógł prawidłowo wykrywać wdech i podawać tlen.
Podczas wdychania powinien być słyszalny przepływ
tlenu do końcówki nosowej.

Uwaga: Jeśli koncentrator tlenu był przez dłuższy okres
przechowywany w warunkach wykraczających poza
normalne zakresy robocze przedstawione w tabeli danych
technicznych, przed włączeniem go należy poczekać,aż osiągnie standardową temperaturę pracy.
Uwaga: Dostawca sprzętu powinien ocenić
zapotrzebowanie na zapasowy tlen na wypadek
awarii koncentratora lub zasilania, a także okresowo,
gdy zmianie ulegają:- stan pacjenta- środowisko, w którym żyje pacjent- możliwość zaopatrzenia pacjenta w zapasowy tlen.

Uwaga: Akumulatory litowe mogą trwale utracić
pojemność, jeśli przy pełnym naładowaniu lub
rozładowaniu będą narażone na ekstremalnie
wysokie temperatury. Jeśli akumulator ma być długo
przechowywany, należy naładować go w 25–50%
i zapewnić warunki temperatur w przedziale od-20°C do 20°C (od -4°F do 68°F).
Uwaga: Nie używać koncentratora tlenu bez założonego
filtra wlotu powietrza. Jeśli dostarczono filtr zastępczy,założyć go przed przystąpieniem do czyszczenia
filtra głównego. Wymyć zanieczyszczony filtr w
ciepłym roztworze mydła w wodzie, a następnie przed
zamontowaniem wysuszyć.
Uwaga: Producent nie zaleca sterylizowania tego sprzętu.
Uwaga: Podczas używania koncentratora tlenu
nie istnieje ryzyko wyczerpania zapasu tlenu
w pomieszczeniu.
OSTRZEŻENIE: NIE NALEŻY PODŁĄCZAĆ
KONCENTRATORA TLENU DO
ZEWNĘTRZNEGO ŹRÓDŁA ZASILANIA(AC/DC) BEZ ZAMONTOWANEGOAKUMULATORA. NIE WŁĄCZAĆ
ANI NIE PODŁĄCZAĆ URZĄDZENIA,
GDY AKUMULATOR JEST ODŁĄCZONY.
USUWAĆ AKUMULATOR TYLKO WTEDY, GDYURZĄDZENIE JEST WYŁĄCZONE I ODŁĄCZONE OD
ŹRÓDŁA ZASILANIA. NIE ODŁĄCZAĆ AKUMULATORA,GDY KONCENTRATOR JEST WŁĄCZONY LUB
PODŁĄCZONY DO ŹRÓDŁA ZASILANIA.

PN MN232-C4 L | Instrukcja obsługi PL - 487

FreeStyle Comfort
Dane techniczne
FreeStyle Comfort
Wymiary (z akumulatorem standardowym)

25,4 x 18,5 x 7,9 cm (10,0 x 7,3 x 3,1 cala)

Wymiary (z akumulatorem powiększonym)

27,9 x 18,5 x 7,9 cm (11,0 x 7,3 x 3,1 cala)

Masa (z akumulatorem standardowym)

2,3 kg (5 funtów)

Masa (z akumulatorem powiększonym)

2,7 kg (6 funtów)

Nominalny poziom hałasu***

39,93 dB(A) ± 0,42 dB(A) przy 2

Ustawienia przepływu

Ustawienie impulsowe 1, 2, 3, 4, 5

Ustawienie podawania O2[(przy 20 odd./min STPD) Wszystkie
wolumeny +/- 15%]Stężenie tlenu*
Czujnik tlenuCiśnienie maksymalne na wylocie
Czułość dozowania
Zasilanie prądem przemiennym (AC)Zasilanie prądem stałym (DC)Rodzaj akumulatora
Pojemność akumulatora

Ustawienie 1 — 210 ml/min
Ustawienie 2 — 420 ml/min
Ustawienie 3 — 630 ml/min
Ustawienie 4 — 840 ml/min
Ustawienie 5 — 1050 ml/min
90% (+5,5% / -3%)
Tak
<30 psi> -0,1 cm H2O (regulowane)
100–240 VAC, 50–60 Hz, maks. 120 W
11–18 V DC (maks. 10 A)
Litowo-jonowy
Pojedyncze ogniwo: 6700 mAh; podwójne ogniwo: 13 400 mAh

Czas ładowania akumulatora

Akumulator standardowy: 3,5 godz.
Akumulator powiększony: 6 godz.

Czasy pracy na akumulatorze

Czas pracy na akumulatorze standardowym: Ustawienie 2 — do 4 godzin
Czas pracy na akumulatorze powiększonym: Ustawienie 2 — do 8 godzin

Dopuszczalna temperatura robocza**

Od 5°C do 40°C (od 41°F do 104°F)

Temperatura podczas transportu
i przechowywania (jednostka)

Od -25°C do +70°C (od -13°F do +158°F)
na wysokości maks. 10 000 stóp (3048 m)

Temperatura podczas transportu
i przechowywania (akumulator)

Od -25°C do +70°C (od -13°F do +158°F)
na wysokości maks. 10 000 stóp (3048 m)

Wysokość robocza**
Dopuszczalna wilgotność podczas pracy
Wilgotność podczas przechowywania
Ograniczona gwarancja-1250–10 000 ft (-381–3048 m) (testowano przy 700–1060 hPa)
Wilgotność względna 15–95% (bez kondensacji)
0–90%, bez kondensacji
Patrz deklaracja gwarancyjna* Przy ciśnieniu atmosferycznym 101 kPa (14,7 psi) w temperaturze 21°C (70°F)** Jednostka będzie działać w każdym środowisku pod warunkiem utrzymania określonych zakresów temperatury
i wilgotności. Jeśli urządzenie będzie używane w warunkach wykraczających poza te wartości robocze, stężenie tlenu
generowane przy wyższych ustawieniach przepływu może być niższe niż w danych technicznych. Ekran przygotowaniamoże być wyświetlany do czasu uzyskania przez koncentrator wymaganego stężenia (maksymalnie 15 minut). W tym
czasie można normalnie użytkować urządzenie.
*** Natężenie dźwięku mierzone metodą testową nr 14-1 10/2018 MDS-Hi.
Przewidywany okres eksploatacji urządzenia to co najmniej pięć lat.

488- PL PN MN232-C4 L | Instrukcja obsługi

FreeStyle Comfort
Instrukcja obsługi
1. Ustaw urządzenie FreeStyle Comfort w miejscu
odpowiednim do użytkowania.
a. Należy zapewnić odpowiednią wentylację izadbać, aby filtry wlotów i wylot powietrza nie
były zablokowane.
b. Koncentrator należy ustawić tak, aby wszystkiewskaźniki oraz alarmy dźwiękowe i graficzne
były dobrze widoczne i słyszalne.
c. Przed uruchomieniem urządzenia upewnij się,że filtry wlotów powietrza są zamontowane.
Jeśli brakuje filtra, skontaktuj się z dostawcą
sprzętu.
2. Podłącz akumulator do koncentratora.
a. Ustaw akumulator centralnie względem dolnejczęści obudowy urządzenia FreeStyle Comfort.

3. Podłącz zasilacz AC lub DC do gniazda
znajdującego się w prawej dolnej części z przoduurządzenia.

4. Podłącz wąsy tlenowe do króćca wylotowego tlenu.
Przed rozpoczęciem użytkowania upewnij się, żepołączenie jest prawidłowe.
UWAGA: Stosowanie wąsów innych niż zalecane możepowodować pogorszenie wydajności.

Uwaga: dla dostawcy sprzętu: Poniżej wyszczególniono
akcesoria do podawania tlenu zalecane do używaniaz koncentratorem tlenu FreeStyle Comfort:b. Wsuwaj akumulator, aż usłyszysz kliknięcie
potwierdzające, że jest on osadzony i zablokowany.

• Wąsy tlenowe z przewodami o długości 2,1 m (7 stóp)(maks. przepływ: 6 l/min): Numer części
CAIRE 5408-SEQ• Dodatkowe informacje o zalecanych akcesoriachdostępne są w witrynie internetowej www.caireinc.com.
5. Włącz koncentrator FreeStyle Comfort, naciskając
przycisk zasilania, i wybierz przepisane ustawienie
przepływu za pomocą przycisków regulacji (+) i (-).

PN MN232-C4 L | Instrukcja obsługi PL - 489

FreeStyle Comfort

Zostanie wyemitowany krótki sygnał dźwiękowy
wskazujący, że urządzenie zostało uruchomione.
Uwaga: Przy pierwszym włączeniu koncentratora
FreeStyle Comfort domyślne ustawienie przepływu to 2.
Jeśli zostanie zmienione, urządzenie zapamięta je przynastępnym uruchomieniu.
6. Załóż wąsy tlenowe na nos zgodnie z dostarczoną
wraz z nimi instrukcją producenta i oddychaj
normalnie. Przy każdym wdechu powinien
być wyczuwalny strumień tlenu dostarczanegoprzez wąsy.
7. Aby zmienić ustawienie przepływu, naciskaj
przyciski regulacji (+) i (-), aż uzyskasz wymaganenatężenie. Różnica w dźwięku podczas zmieniania
ustawień jest normalna.

Częstotliwość
oddechów
15
20
25
30
35
40

Ustawienie objętości impulsu bez funkcji
autoSat (ml)*
1
2
3
4
5
14,0
10,5
8,4
7,0
6,0
5,3

28,0
21,0
16,8
14,0
12,0
10,5

42,0
31,5
25,2
21,0
18,0
15,8

56,0
42,0
33,6
28,0
24,0
21,0

70,0
52,5
42,0
35,0
30,0
26,3

8. Wyłącz koncentrator FreeStyle Comfort, naciskając
przycisk zasilania.
Uwaga: Wyświetlacz przełączy się w tryb gotowości i zostanie przygaszony po 20 sekundach. Nacisnąć dowolny
przycisk, aby wybudzić wyświetlacz z trybu gotowości.
9. Funkcje zaawansowane, patrz instrukcje poruszania
się po menu.

SPECJALNE FUNKCJE PRODUKTU
autoSAT®
Zastrzeżona funkcja firmy CAIRE — autoSAT —umożliwia koncentratorowi FreeStyle Comfort
dostarczanie stałej objętości tlenu (bolusa) na każdy
oddech, do maksymalnej szybkości oddechu.
Ponieważ szybkość oddechu użytkownika wzrasta
podczas czynności takich jak chodzenie, funkcja
autoSAT dostosowuje działanie urządzenia, aby
utrzymać ustawioną objętość bolusa.
UltraSense®
Ten czuły regulator dawkowania wykrywa zmianę
ciśnienia podczas oddychania i uwalnia impuls tlenu
przy każdym wdechu.

Częstotliwość
oddechów
15
20
25
30
35
40

Ustawienie objętości impulsu autoSat (ml)*
1
2
3
4
5
10,5
10,5
10,5
10,5
10,5
10,5

21,0
21,0
21,0
21,0
21,0
21,0

31,5
31,5
31,5
31,5
30,0
26,3

42,0
42,0
42,0
35,0
30,0
26,3

*Tolerancja wynosi +/-15%

490- PL PN MN232-C4 L | Instrukcja obsługi

52,5
52,5
42,0
35,0
30,0
26,3

autoDOSE®
Jeśli wystąpi alarm „brak oddechu”, funkcja tabędzie impulsowo podawała tlen z częstotliwością
20 oddechów na minutę, przy bieżącym ustawieniu
przepływu, aż do zlikwidowania alarmu.

FreeStyle Comfort
Zasilanie
Koncentrator FreeStyle Comfort może być zasilanyna trzy sposoby:• Akumulator• Zasilanie prądem przemiennym (AC)• Zasilanie prądem stałym (DC)Informacje na temat działania każdego z tych
komponentów dostępne są poniżej.
OSTRZEŻENIE: KORZYSTAĆ WYŁĄCZNIEZ ZASILACZY WSKAZANYCH W TEJINSTRUKCJI OBSŁUGI. STOSOWANIEZASILACZY I AKUMULATORÓW FIRM
INNYCH NIŻ CAIRE MOŻE SPOWODOWAĆUSZKODZENIE I NIEPRAWIDŁOWE
DZIAŁANIE URZĄDZENIA.

ODŁĄCZANIE AKUMULATORA

Nacisnąć przycisk zwalniania akumulatora w dół,w kierunku przeciwnym do urządzenia FreeStyle
Comfort. TRZYMAJĄC ten przycisk, wysuń
akumulator z koncentratora.
OSTRZEŻENIE: PRZED ODŁĄCZENIEM
AKUMULATORA KONCENTRATOR TLENU
NALEŻY BEZWZGLĘDNIE WYŁĄCZYĆ
I ODŁĄCZYĆ OD ZEWNĘTRZNEGOŹRÓDŁA ZASILANIA.

AKUMULATOR

Urządzenie FreeStyle Comfort może być zasilane
przez dostarczany wraz z nim akumulator.
Funkcjonalność koncentratora nie jest w żaden
sposób ograniczona podczas pracy na akumulatorze.
Przy zasilaniu z akumulatora dostępne są wszystkie
ustawienia natężenia przepływu.

SPRAWDZANIE STANU NAŁADOWANIAAKUMULATORA
PODŁĄCZANIE AKUMULATORA

Ustaw akumulator centralnie względem dolnej części
obudowy urządzenia FreeStyle Comfort. Wsuńakumulator, aż usłyszysz kliknięcie potwierdzające, żezatrzasnął się na miejscu.

Stan naładowania akumulatora, gdy jest on podłączony
do urządzenia FreeStyle Comfort, można sprawdzić,
spoglądając na ikonę baterii i szacowny pozostały czas
pracy prezentowany na wyświetlaczu.
80–100%60–80%

40–60%

20–40%

20%

PN MN232-C4 L | Instrukcja obsługi PL - 491

FreeStyle Comfort
Uwaga: Jeśli podłączony jest zasilacz, zamiast procentowego poziomu naładowania wyświetlany będzie wskaźnik
ładowania.
Aby sprawdzić poziom naładowania akumulatora,
gdy nie jest podłączony do koncentratora FreeStyle
Comfort, połóż go na płaskiej powierzchni i nacisnąć
przycisk TEST. Po jego prawej stronie zaświecą się
kontrolki akumulatora (25–100%), wskazując poziom
naładowania ogniw.

CZAR PRACY NA AKUMULATORZE

Gdy zasilacz AC lub DC zostanie odłączony,koncentrator automatycznie przełączy się na zasilaniez akumulatora, jeśli jest on zainstalowany. W tabeliponiżej zaprezentowano typowe czasy pracy dla
nowego akumulatora. Częstotliwość oddechu nie mawpływu na czas pracy na akumulatorze.
Ustawienie
impulsów

Akumulator
standardowy(8 ogniw)

Akumulatorpowiększony(16 ogniw)

1

do 8 godzin

do 16 godzin

2

do 4 godzin

do 8 godzin

3

do 3 godzin

do 6 godzin

4

do 2,25 godz.

do 4,5 godz.

5

do 2 godzin

do 4 godzin

TYPOWY CZAS ŁADOWANIAAKUMULATORAPIERWSZE ŁADOWANIE

Nowy akumulator dostarczany wraz z urządzeniem
FreeStyle Comfort jest tylko częściowo naładowany
w chwili opuszczania fabryki. Przed pierwszym
użyciem należy go całkowicie naładować za pomocą
zasilacza AC lub DC.

INFORMACJE OGÓLNE NA TEMATŁADOWANIA

Akumulator jest ładowany zawsze, gdy jest
zainstalowany w koncentratorze FreeStyle Comfortpodłączonym do zewnętrznego źródła zasilania AC lub
DC (z wyjątkiem samolotu).
Akumulator jest w pełni naładowany, gdy wskaźnik
pokazuje 100%.
Można go też ładować za pomocą opcjonalnejładowarki biurkowej.

492- PL PN MN232-C4 L | Instrukcja obsługi

Zazwyczaj całkowite naładowanie w pełni
rozładowanego akumulatora zajmuje 3,5 godziny
dla modelu standardowego i 6 godzin w przypadku
powiększonego.
Uwaga: Jeśli akumulator został całkowicie rozładowany,
jego ładowanie może nie rozpocząć się od razu.

OKRES EKSPLOATACJI AKUMULATORA

Akumulatory FreeStyle Comfort wytrzymują 300 cykliładowania/rozładowania.

FreeStyle Comfort
Zasilacz AC i DCUrządzenie FreeStyle Comfort dostarczane jest
standardowo z zasilaczami AC i DC.
Na zasilacz AC składają się:
	a. Zasilacz właściwy w obudowie, pracujący
w zakresie napięć 100–240 V i częstotliwości
50–60 Hz.
	b. Przewód zasilania AC podłączany dozasilacza i gniazda prądu przemiennego.

Zasilacz DC składa się z przewodu, który podłącza się
bezpośrednio do urządzenia i gniazda prądu stałego.

ZASILANIE PRĄDEM PRZEMIENNYM (AC)Z zasilania AC można korzystać w domu, gdydostępne jest standardowe gniazdo sieciowe prądu
przemiennego. Koncentrator FreeStyle Comfort
zasilany prądem przemiennym jest w pełni
funkcjonalny. Urządzenie FreeStyle Comfort ładuje
akumulator zawsze, gdy jest podłączone do źródła
zasilania AC (z wyjątkiem samolotu). Ładowanie
odbywa się bez względu na to, czy koncentrator
FreeStyle Comfort jest włączony czy wyłączony.

OSTRZEŻENIE: NIE PODŁĄCZAĆ DOZEWNĘTRZNEGO ŹRÓDŁA ZASILANIA,GDY AKUMULATOR NIE JESTZAINSTALOWANY W KONCENTRATORZETLENU.
Aby podłączyć urządzenie FreeStyle Comfort doźródła zasilania AC:
1. Podłącz zasilacz AC do złącza zasilania w prawej
dolnej części, z przodu koncentratora FreeStyle
Comfort. Upewnij się, że wtyczka jest prawidłowowciśnięta i podłączona.

2. Podłącz przewód zasilania AC do zasilacza prądu
przemiennego.

3. Podłącz drugi koniec przewodu zasilania AC dogniazda ściennego lub innego.

PN MN232-C4 L | Instrukcja obsługi PL - 493

FreeStyle Comfort
4. Włącz koncentrator FreeStyle Comfort, naciskając
przycisk zasilania, i wybierz przepisane ustawienie
przepływu za pomocą przycisków regulacji (+) i (-).

4. Włącz koncentrator FreeStyle Comfort, naciskając
przycisk zasilania, i wybierz przepisane ustawienie
przepływu za pomocą przycisków regulacji (+) i (-).
Uwaga: Jeśli konieczne jest naładowanie akumulatora,
na ikonie baterii na wyświetlaczu widoczny będzie jeden
pomarańczowy pasek.

ZASILANIE PRĄDEM STAŁYM (DC)

Zasilacz DC służy do podłączania do gniazd
w samochodach, łodziach i innych pojazdach.
Koncentrator FreeStyle Comfort zasilany prądem
stałym jest w pełni funkcjonalny. W razie potrzebyurządzenie FreeStyle Comfort ładuje akumulator, gdy
jest podłączone do źródła zasilania DC. Ładowanie
odbywa się bez względu na to, czy urządzenie jestwłączone czy wyłączone.
OSTRZEŻENIE: NIE PODŁĄCZAĆ DOZEWNĘTRZNEGO ŹRÓDŁA ZASILANIA,JEŻELI AKUMULATOR NIE JESTZAINSTALOWANY W KONCENTRATORZETLENU.
1. Uruchom pojazd.
2. Podłącz zasilacz DC do gniazda zasilania w prawej
dolnej części, z przodu koncentratora FreeStyle
Comfort. Upewnij się, że wtyczka jest prawidłowowciśnięta i podłączona.

3. Podłącz drugi koniec przewodu zasilania DC dogniazda zasilania w samochodzie.

494- PL PN MN232-C4 L | Instrukcja obsługi

PRZESTROGA: Aby zapobiec
ewentualnemu pożarowi lub przegrzewaniu
się urządzenia, należy upewnić się,że gniazdo 12 V DC jest wolne od
zanieczyszczeń, a wtyczka adaptera
zasilacza DC została prawidłowo
i bezpiecznie włożona do gniazda.
PRZESTROGA: Powierzchnie metalowe zasilacza
DC w trakcie korzystania z niego mogą być BARDZOGORĄCE ze względu na prąd. Przed dotknięciem
poczekać na ostygnięcie powierzchni.
PRZESTROGA: Nie używać zasilacza DC z żadnymi
przedłużaczami, rozgałęziaczami ani adapterami.
Stosowanie niezatwierdzonych akcesoriów możespowodować ich przegrzanie lub uszkodzenie.
PRZESTROGA: Podłączenie urządzenia do gniazda
12 V DC może doprowadzić do wyczerpania
akumulatora pojazdu w przypadku niepracującego
silnika, nawet jeśli urządzenie jest wyłączone.
PRZESTROGA: Nie wolno uruchamiać pojazdu
podczas podłączania zasilania prądem stałym.

FreeStyle Comfort
Czyszczenie, pielęgnacja i rutynowe
czynności konserwacyjne
RUTYNOWE CZYNNOŚCIKONSERWACYJNE

Konserwacja zapobiegawczaUrządzenie nie wymaga konserwacji. Filtry znajdujące
się wewnątrz urządzenia mają wystarczyć na całyokres użytkowania urządzenia. Jeżeli urządzenie nie
działa prawidłowo, należy je zwrócić dostawcy w celu
naprawy. Serwisowanie podzespołów znajdujących sięwewnątrz obudowy koncentratora FreeStyle Comfortmoże być wykonywane tylko przez dostawcę sprzętu
do podawania tlenu.
Operator urządzenia odpowiada za wykonywanie niżej
wymienionej procedury konserwacyjnej:• Raz na tydzień — wyczyścić filtry wlotów powietrza.
OSTRZEŻENIE: NIE OTWIERAĆ ANI NIE
ZDEJMOWAĆ PLASTIKOWYCH POKRYW
KONCENTRATORA TLENU. MOŻE TOROBIĆ JEDYNIE WYKWALIFIKOWANYSERWISANT.

FILTRY WLOTÓW POWIETRZA

Powietrze doprowadzane jest do koncentratora
FreeStyle Comfort przez filtry wlotów powietrzaznajdujące się z przodu urządzenia na dole. Filtry tezapobiegają dostawaniu się do urządzenia kurzu i
innych dużych cząstek znajdujących się w powietrzu.
Należy je czyścić co najmniej raz w tygodniu.
Dostawca sprzętu do podawania tlenu może zalecaćczęstsze czyszczenie w zależności od warunków
eksploatacji. Aby odpowiednio wyczyścić filtry na
wlotach powietrza, należy wykonać następujące
czynności:1. Zdjąć dwa filtry z przodu urządzenia FreeStyle
Comfort, naciskając odpowiednie uszko w góręi pociągając na zewnątrz.

OSTRZEŻENIE: NIE UŻYWAĆ KONCENTRATORA TLENU BEZ ZAŁOŻONEGOFILTRA WLOTU POWIETRZA.

KONSERWACJA AKUMULATORA

Nie narażać akumulatora na działanie płynów.
Nie rozbierać go ani nie otwieraj.
Nie narażać na działanie otwartego ognia.
Nie upuszczać.
Trzymać z dala od dzieci.
Jeśli na akumulatorze widoczne są uszkodzenia,
przed użyciem skontaktować się z dostawcą sprzętu
do podawania tlenu.
Nie umieszczać akumulatora wewnątrz lub w
pobliżu kuchenki mikrofalowej ani innych urządzeń
kuchennych.

PRZECHOWYWANIE AKUMULATORA

Akumulatora koncentratora tlenu nie trzeba całkowicie
rozładowywać przed ładowaniem. Zaleca się ładować
go po każdym użyciu urządzenia.
Nie zostawiać akumulatora w miejscach, w których
narażony byłby na działanie wysokiej temperatury, na
przykład silnie nasłonecznionych lub w samochodzie.
W przeciwnym razie może dojść do pogorszenia jegowydajności i skrócenia czasu eksploatacji. Możeteż dojść do nagrzania akumulatora, wydostawania
się z niego dymu, zapłonu, a nawet wybuchu.
Akumulator należy przechowywać w miejscu o
ustabilizowanych warunkach i niskiej wilgotności
(poniżej 70% wilgotności względnej), wolnym od
gazów korozyjnych. Aby zapobiec rdzewieniu, unikać
warunków, w których dochodzi do kondensacji, np.
nagłych zmian temperatury otoczenia.
Jeśli akumulator będzie przechowywany przez miesiąc
lub dłużej, zaleca się rozładować go lub naładowaćczęściowo do poziomu od 50% do 75% (od dwóch dotrzech diod LED). Po naładowaniu odłącz akumulator
od koncentratora tlenu.

2. Wymyć oba filtry w ciepłym roztworze mydła
w wodzie.
3. Dokładnie wypłukać filtry i wytrzeć miękką,niepozostawiającą włókien ściereczką. Przed
ponownym założeniem upewnić się, że filtry są
całkowicie suche.
4. Zamontować czyste i suche filtry.
PN MN232-C4 L | Instrukcja obsługi PL - 495

FreeStyle Comfort
CZYSZCZENIE

Przed czyszczeniem WYŁĄCZYĆ urządzenie
i odłączyć je od zasilania. NIE spryskiwać obudowyzewnętrznej. Stosować wilgotną (nie mokrą)szmatkę lub gąbkę. Spryskać ściereczkę lub gąbkę
łagodnym roztworem środka dezynfekującego, aby
wyczyścić obudowę i zasilacze. Postępować zgodnie
z instrukcjami producenta środka czyszczącego.
OSTRZEŻENIE: ABY UNIKNĄĆ
PRZYPADKOWEGO PORAŻENIA
PRĄDEM LUB POPARZENIA, PRZEDPRZYSTĄPIENIEM DO CZYSZCZENIAURZĄDZENIA NALEŻY JE WYŁĄCZYĆ
I WYJĄĆ PRZEWÓD ZASILANIA ZGNIAZDA. SERWISOWAĆ KONCENTRATORI ZDEJMOWAĆ POKRYWY MOŻEWYŁĄCZNIE DOSTAWCA SPRZĘTU LUB
WYKWALIFIKOWANY TECHNIK.
OSTRZEŻENIE: UWAŻAĆ, ABY NIE ZAMOCZYĆ
KONCENTRATORA TLENU I ABY DO WEWNĄTRZ
NIE DOSTAŁ SIĘ ŻADEN PŁYN. MOGŁOBY TO
SPOWODOWAĆ USZKODZENIE LUB WYŁĄCZENIEURZĄDZENIA I ZWIĘKSZYĆ RYZYKO PORAŻENIA
PRĄDEM I POPARZEŃ.
OSTRZEŻENIE: NIE STOSOWAĆ PŁYNÓWBEZPOŚREDNIO NA URZĄDZENIU. NIEKTÓRE
ŚRODKI CHEMICZNE, KTÓRYCH NIE NALEŻYSTOSOWAĆ: ALKOHOL I PRODUKTY NA BAZIEALKOHOLU, KONCENTRATY NA BAZIE CHLORU(CHLOREK ETYLENU) ORAZ PRODUKTY NA BAZIEOLEJU. ŚRODKÓW TYCH NIE NALEŻY UŻYWAĆ
DO CZYSZCZENIA PLASTIKOWEJ OBUDOWYURZĄDZENIA, PONIEWAŻ MOGĄ JE USZKODZIĆ.
OSTRZEŻENIE: DO CZYSZCZENIA OBUDOWY,
PANELU STEROWANIA I PRZEWODU ZASILANIA
UŻYWAĆ WYŁĄCZNIE ŁAGODNEGO DOMOWEGOŚRODKA CZYSZCZĄCEGO I WILGOTNEJ SZMATKI(NIE MOKREJ) LUB GĄBKI. PO CZYSZCZENIUWYTRZEĆ DO SUCHA. UWAŻAĆ, ABY ŻADEN PŁYNNIE DOSTAŁ SIĘ DO WNĘTRZA URZĄDZENIA.
SZCZEGÓLNĄ UWAGĘ ZWRÓCIĆ NA ZŁĄCZE
WYLOTOWE TLENU SŁUŻĄCE DO PODŁĄCZANIAWĄSÓW — UPEWNIĆ SIĘ, ŻE NIE MA NA NIM KURZU,WODY ANI INNYCH ZANIECZYSZCZEŃ.

496- PL PN MN232-C4 L | Instrukcja obsługi

OSTRZEŻENIE: DO CZYSZCZENIA AKUMULATORA
UŻYWAĆ WILGOTNEJ (NIE MOKREJ) SZMATKI
LUB GĄBKI. SPRYSKAĆ SZMATKĘ LUB GĄBKĘ
ŁAGODNYM DOMOWYM ŚRODKIEM CZYSZCZĄCYMI WYTRZEĆ OBUDOWĘ AKUMULATORA.
OSTRZEŻENIE: NIE CZYŚCIĆ URZĄDZENIAZ UŻYCIEM MYJKI AUTOMATYCZNEJ.

WYMIANA WĄSÓW

Bezwzględnie przestrzegać instrukcji prawidłowegoużytkowania dostarczonej przez producenta wąsów.
Wymieniać wąsy tlenowe lub przewody tlenowe co
3-6 miesięcy (jeśli używane są codziennie, co 3 miesiące)zgodnie z zaleceniami producenta urządzenia, producenta
komponentu lub dostawcy tlenu. Instrukcje czyszczenia
i wymiany powinien dostarczyć lekarz lub dostawcasprzętu do podawania tlenu.
Części zamienne do urządzenia dostępne są u
dostawcy sprzętu do podawania tlenu.

FreeStyle Comfort
Stany alarmowe
Nr priorytetu

Opis alarmu

1

Usterka systemu — wysoka
temperatura
akumulatoraWyświetlacz

Sygnał
dźwiękowy

Dodatkowe informacje / sugerowane
działanie

1 sygnałdźwiękowy
co 20 sPo obniżeniu temperatury akumulatora dowartości granicznej powodującej wyłączenie
awaryjne urządzenie wygeneruje alarm.
Jeśli limit zostanie osiągnięty, urządzenie sięwyłączy. Jeśli będzie podłączone zewnętrzneźródło zasilania, przez 2 minuty będzie
wyświetlany alarm na ekranie oraz emitowany
sygnał dźwiękowy.
Wykonać test, używając akumulatora w temperaturze pokojowej. Jeśli problem nie ustąpi,skontaktować się z dostawcą.

2

Usterkasystemu —wysokatemperatura

1 sygnałdźwiękowy
co 20 sUrządzenie wyemituje alarm, gdy temperaturawewnętrzna zbliży się do limitu powodującego
wyłączenie.
Jeśli limit zostanie osiągnięty, urządzenie sięwyłączy. Przez 2 minuty będzie wyświetlanyalarm na ekranie oraz emitowany sygnałdźwiękowy.
Upewnić się, że nie ma żadnych niedrożności
na wlotach i wylotach powietrza urządzenia.
Jeśli problem nie ustąpi, skontaktować się z
dostawcą.

3

Usterkasystemu —wysokieciśnienie (błądwysokiegociśnienia
systemu lubsprężarki)

1 sygnałdźwiękowy
co 20 s4

Usterkasystemu —natężenie
przepływu
(awariazaworu OCD,
niedrożność
linii gazu)

1 sygnałdźwiękowy
co 30 sJeśli problem nie ustąpi, skontaktować sięz dostawcą.

5Wyłączenie z
powodu niskiego poziomu
naładowania
akumulatora

1 sygnałdźwiękowy
co 30 sAlarm emitowany do momentu wyczerpania
akumulatora i wyłączenia urządzenia.

Jeśli limit zostanie osiągnięty, urządzenie sięwyłączy. Przez 2 minuty będzie wyświetlanyalarm na ekranie oraz emitowany sygnałdźwiękowy.
Jeśli problem nie ustąpi, skontaktować sięz dostawcą.

Naładować akumulator.

PN MN232-C4 L | Instrukcja obsługi PL - 497

FreeStyle Comfort
Nr priorytetu

Opis alarmu

6

Nie wykryto
oddechuWyświetlacz

Sygnał
dźwiękowy

Dodatkowe informacje

1 sygnałdźwiękowy
co 10 minut

Nie wykryto oddechu przez 15 sekund.
Aktywacja po pierwszym oddechu lub wyłączeniu ekranu przygotowania, w zależności
od tego, co nastąpi szybciej.
Sprawdzić wąsy pod kątem zagięć i
przecieków. Zwiększyć ustawienie czułości.
Jeśli problem nie ustąpi, skontaktowaćsię z dostawcą.

7

Nagrzewanie

BrakUrządzenie uruchomi się i będzie działało
w trybie automatycznego dozowania do
czasu wykrycia wdechu lub zakończenia
przygotowania, w zależności od tego,
co nastąpi szybciej.

8

Niskiestężenie

1 sygnałdźwiękowy
co 10 minut,przez maksymalnie
30 minutJeśli problem nie ustąpi, skontaktować sięz dostawcą.

9

Ostrzeżenie
o niskim
poziomienaładowania
akumulatora

1 sygnałdźwiękowy,
bez powtarzania

Naładować akumulator.

10

Zbyt duża
częstość
oddechu

1 sygnałdźwiękowy,
bez powtarzaniaCzęstotliwość oddechu przekracza
40 oddechów/min.

11

Usterkasystemu —uszkodzenie
wentylatora

BrakJeśli problem nie ustąpi, skontaktować sięz dostawcą.

12

Usterkasystemu —niskieciśnienie

BrakJeśli problem nie ustąpi, skontaktować sięz dostawcą.

13

Usterkasystemu —niska
temperatura

Brak

Pozwolić, aby urządzenie ogrzało się dotemperatury pokojowej. Jeśli problem nieustąpi, skontaktować się z dostawcą.

498- PL PN MN232-C4 L | Instrukcja obsługiJeśli dawka jest nieprawidłowo wyzwalana,należy zmniejszyć ustawienie czułości.
Jeśli problem nie ustąpi, skontaktować sięz dostawcą.

FreeStyle Comfort
Nr priorytetu

Opis alarmu

Info

UruchomienieWyświetlacz

Sygnał
dźwiękowy

Dodatkowe informacje

1 sygnałinformacyjny

Informacja na wyświetlaczu po uruchomieniu
oraz pojedynczy sygnał dźwiękowy potwierdzający sprawność brzęczyka.
Na ekranie w prawym dolnym rogu wyświetlana jest bieżąca wersja oprogramowania
układowego.

Info

Normalnyekran roboczy
w trybie autoSAT (zasilanie
DC niepodłączone)

Brak

Info

Normalnyekran roboczy
przy wyłączonej funkcji
autoSAT(zasilanie DC
niepodłączone)

Brak

Info

Zmianaustawienia

1 sygnałinformacyjny

InfoPodłączyćzewnętrzne źródło
zasilania przywyłączonymurządzeniu

Brak

Info

Podłączyćzewnętrzneźródłozasilania przywłączonymurządzeniu

Brak

Info

Odłączyćzewnętrzneźródłozasilania przywyłączonymurządzeniu

1 sygnałinformacyjnyWyświetla ekran CAIRE przez 2 sekundy,następnie stan baterii i zasilania przez 5 s,
po czym wyświetlacz się wyłącza.

Wyświetla ekran CAIRE przez 2 sekundy,po czym wyświetlacz się wyłącza.

PN MN232-C4 L | Instrukcja obsługi PL - 499

FreeStyle Comfort
Nr priorytetu

Opis alarmuWyświetlacz

InfoOdłączyćzewnętrzneźródłozasilania przywłączonymurządzeniu(z zainstalowanym
akumulatorem)

1 sygnałinformacyjny

Info

W trybie gotowości — gdynaciśnięty zostanie przycisk
w górę lub w
dół, na ekraniewyświetlony
zostanie stan
akumulatorai ładowania

Brak

Info

Tryb
samolotowy
przy zasianiu
z akumulatora

1 długisygnał

Info

Trybsamolotowy
przy zasilaniu
sieciowym

1 długisygnał

500- PL PN MN232-C4 L | Instrukcja obsługi

Sygnał
dźwiękowy

Dodatkowe informacjeWyświetlacz włączy się na 5 s, a następnie
wyłączy.

FreeStyle Comfort
Instrukcje poruszania się po menu
Do menu można przejść w trybie gotowości lub w
trybie działania. Jeśli w trybie działania wystąpi alarm,menu zostanie automatycznie zamknięte. Jeśli przez
20 sekund nie będą naciskane żadne przyciski, menu
zostanie automatycznie zamknięte. W trybie gotowości
może być konieczne dwukrotne naciśnięcie przycisku
Menu w celu wybudzenia urządzenia, aby możliwe
było przejście do menu.
Dostępne są cztery ekrany menu:
1. Ekran stanu: wyświetla wersję oprogramowaniasprzętowego, czas pracy, liczbę cykli ładowania
akumulatora

Tryb samolotowy

2. Tryb samolotowy (WŁ./WYŁ.)3. Czułość: regulowana (1–5)
4. Funkcja autoSAT (WŁ./WYŁ.)Czułość

Przykład ekranu stanuDZIAŁANIE MENU USTAWIEŃ
Aby przejść do menu ustawień, należy nacisnąć
przycisk Menu/Select (Menu/Wybór)
działania lub gotowości.
Do przewijania menu służą przyciski

autoSAT

w trybie.

Strona stanu

PN MN232-C4 L | Instrukcja obsługi PL - 501

FreeStyle Comfort
STRONA STANU

Aby anulować włączenie trybu samolotowego, nacisnąć

Strona stanu koncentratora tlenu FreeStyle Comfortwyświetla wersję oprogramowania sprzętowego,czas pracy i liczbę cykli baterii na podłączonym
akumulatorze.

, gdy widoczna jest ikona CANCEL (ANULUJ).

Potwierdzone włączenie trybu samolotowego

Strona stanu

Aby wyłączyć tryb samolotowy, powtórzyć powyższe
kroki, przestawiając stan na pozycję OFF (WYŁ.).

Czas pracy jest przedstawiany w pełnych godzinach.
Liczba rozładowań akumulatora to liczba cykli,
w których dołączony akumulator został wyczerpany
i ponownie naładowany.

Naciśnięcie przycisku, gdy na ekranie jest
widoczny symbol „Dalej”, powoduje przejście
do następnego ekranu opcji.

TRYB SAMOLOTOWY

Tryb samolotowy jest przeznaczony do użycia w
trybie pracy. Tryb samolotowy zostanie automatyczniewyłączony, gdy urządzenie zostanie pozostawione
w trybie gotowości na zasilaniu akumulatorowym.

Koncentrator FreeStyle Comfort jest dostarczanyz wyłączonym trybem samolotowym.

Uwaga: Więcej informacji na temat trybu samolotowego
zawiera rozdział Włączanie trybu samolotowego.

TRYB USTAWIANIA CZUŁOŚCI

Ekran trybu samolotowego

Koncentrator tlenu FreeStyle Comfort jest dostarczanyz fabrycznie ustawionym trybem czułości 2 i w
normalnych warunkach nie powinien wymagać
regulacji. Ustawienie czułości można zmieniać w
zakresie 1–5, gdzie 1 to najmniejsza, a 5 największa
wartość czułości.

Aby ten tryb włączyć, nacisnąć
, gdy widoczna
jest ikona CHANGE (ZMIEŃ). Wartość zmieni się
na niebieskie słowo „ON” (WŁ.).

Ekran trybu ustawiania czułości z wartością 2

Wybrane włączenie trybu samolotowego

Aby potwierdzić włączenie trybu samolotowego,nacisnąć
, gdy widoczna jest ikona SAVE(ZAPISZ). Wartość zmieni się na czarne słowo„ON” (WŁ.).

502- PL PN MN232-C4 L | Instrukcja obsługi

Aby zmienić czułość, nacisnąć
, gdy widoczna
jest ikona CHANGE (ZMIEŃ). Wartość zmieni się na
kolejną i zostanie wyświetlona na niebiesko.

FreeStyle Comfort
TRYB AUTOSAT
Koncentrator FreeStyle Comfort jest dostarczanyz włączonym trybem autoSAT.

Wybrane ustawienie czułości 3

Aby potwierdzić wybrane ustawianie czułości 3, nacisnąć
, gdy widoczna jest ikona SAVE (ZAPISZ).
Wartość zmieni się na „3” i zostanie wyświetlona
na czarno.
Aby anulować ustawianie czułości, nacisnąć
,gdy widoczna jest ikona CANCEL (ANULUJ).

Potwierdzone ustawienie czułości 3

Ekran główny trybu autoSAT

Aby ten tryb wyłączyć, nacisnąć
, gdy widoczna
jest ikona CHANGE (ZMIEŃ). Wartość zmieni się
na niebieskie słowo „OFF” (WYŁ.).

Wybrane wyłączenie trybu autoSAT

Aby potwierdzić wyłączenie trybu autoSAT, nacisnąć
, gdy widoczna jest ikona SAVE (ZAPISZ).
Wartość zmieni się na czarne słowo „OFF” (WYŁ.).
Aby anulować włączenie trybu autoSAT, nacisnąć
gdy widoczna jest ikona CANCEL (ANULUJ).

,

Potwierdzone ustawienie czułości 5 (maksymalne)Naciśnięcie przycisku, gdy na ekranie jest
widoczny symbol „Dalej”, powoduje przejście
do następnego ekranu opcji.

Potwierdzone wyłączenie trybu autoSAT

Aby włączyć tryb autoSAT, powtórzyć powyższe
kroki, przestawiając stan na pozycję ON (WŁ.).
Naciśnięcie przycisku, gdy na ekranie jest
widoczny symbol „Dalej”, powoduje przejście
do następnego ekranu opcji.

PN MN232-C4 L | Instrukcja obsługi PL - 503

FreeStyle Comfort
Rozwiązywanie problemów
ProblemMożliwa przyczyna

RozwiązanieUrządzenie FreeStyle Comfort
pokazuje alarm i emituje
przerywany sygnał dźwiękowy.

Sprawdzić w tabeli Stany
alarmowe.

Sprawdzić w tabeli Stany alarmowe.

Koncentrator tlenu
nie włącza się.

Akumulator jest rozładowany.

Podłączyć urządzenie do gniazda AC lub DC.

Niepodłączony do zewnętrznegoźródła zasilania.

Upewnić się, że połączenia zewnętrzne są
bezpieczne.

Awaria ogólna.

Wykonać pełny cykl zasilania, wyjmując akumulator
i odłączając wszelkie zewnętrzne źródła zasilania
AC lub DC. Następnie ponownie zamontować
akumulator — cykl zasilania zostanie zakończony.
W razie potrzeby ponownie podłączyć zewnętrzneźródło zasilania.
Skontaktować się z dostawcą sprzętu i w razie
potrzeby skorzystać z innego źródła tlenu.

Wszystkie inne problemy.

Wykonać pełny cykl zasilania, wyjmując akumulator
i odłączając wszelkie zewnętrzne źródła zasilania
AC lub DC. Następnie ponownie zamontować
akumulator — cykl zasilania zostanie zakończony.
W razie potrzeby ponownie podłączyć zewnętrzneźródło zasilania.
Skorzystać z innego źródła tlenu, jeśli jestdostępne, i skontaktować się z dostawcą sprzętu.

Uwaga: Niektóre alarmy powodują wyłączenie urządzenia.
Uwaga: W przypadku całkowitej utraty zasilania system alarmowy jest resetowany (czyszczone są wszelkie wcześniejsze
alarmy). Po przywróceniu zasilania wyświetlane są tylko te alarmy, których warunki wciąż są spełnione.

504- PL PN MN232-C4 L | Instrukcja obsługi

FreeStyle Comfort
Informacje dla podróżujących
Koncentrator FreeStyle Comfort spełnia zalecenia
Federalnej Administracji Lotnictwa w zakresieużytkowania urządzeń na pokładzie samolotów
cywilnych. Przed każdą podróżą (samolotem,pociągiem, statkiem itd.) należy:• Upewnić się, że koncentrator jest w dobrym stanie
i działa.
• Skontaktować się z przewoźnikiem co najmniej
48 godzin przed wyruszeniem i poinformować
go o zamiarze podróżowania z urządzeniem
FreeStyle Comfort.
Podczas podróży zaleca się zawsze mieć przysobie zapasowe akumulatory. W przypadku
podróży lotniczych należy zadbać, aby energii z
akumulatorów wystarczyło też na czas spędzony na
lotnisku (odprawa, kontrole bezpieczeństwa itp.)
oraz przerwy w podróży. Większość linii lotniczych
wymaga, aby pasażer miał ze sobą taką liczbę w pełni
naładowanych akumulatorów, aby zapewnić zasilanieurządzenia przez co najmniej 150% przewidywanego
maksymalnego czasu podróży (czasy przelotów,
pobyt na lotnisku przed lotem i po, połączenia
i nieoczekiwane opóźnienia).
Uwaga: W przypadku lotu zaleca się, aby dodatkowe
akumulatory przewozić w bezpiecznej torbie.

PRZEPISY FEDERALNEJ ADMINISTRACJILOTNICTWA

1. Włączyć przenośny koncentrator tlenu
z zainstalowanym akumulatorem i bez zasilaniaze źródła zewnętrznego.

2. Włączyć tryb samolotowy, używając jednej znastępujących metod:
a. Wykonać czynności mające na celu włączenie
trybu samolotowego, opisane w sekcji Instrukcje
poruszania się po menu.

b. Naciśnąć i przytrzymać przyciski „w górę” i„w dół” przez 5 sekund, aż wyemitowany zostanie
długi sygnał dźwiękowy, a na wyświetlaczu pojawi
się symbol samolotu.

Akumulatory zapasowe należy oddzielniezabezpieczyć przed zwarciem, umieszczając jew oryginalnym opakowaniu lub inaczej odizolowując
styki, np. oklejając je taśmą lub umieszczając każdy
akumulator w oddzielnej torebce foliowej lub
woreczku ochronnym. Akumulatory można wnosić
tylko w bagażu podręcznym.

WŁĄCZANIE TRYBU SAMOLOTOWEGO
Zasilanie akumulatorowe
Przenośnego koncentratora tlenu można używać
w samolocie w normalny sposób, gdy jest zasilany
TYLKO z akumulatora, a zewnętrzne źródło zasilania
nie jest podłączone.
Zasilanie zewnętrzneJeśli w samolocie dostępne jest działające gniazdo
elektryczne, z którego można zasilać przenośny
koncentrator tlenu, może ono stanowić alternatywne
źródło zasilania. W takim wypadku, aby przełączyć
urządzenia z zasilania akumulatorowego nazewnętrzne, należy postępować zgodnie z następującą
procedurą:3. Po pojawieniu się symbolu podłączyć przenośny
koncentrator tlenu do gniazda zasilania.

Uwaga: Ze względu na ograniczenia obowiązujące
w samolocie akumulatory nie ładują się w trybie
samolotowym.

PN MN232-C4 L | Instrukcja obsługi PL - 505

FreeStyle Comfort
AkcesoriaWKŁADANIE URZĄDZENIA DO TORBY

Aby zapewnić odpowiednie działanie i bezpieczeństwo,należy używać wyłącznie wyszczególnionych
akcesoriów dostarczanych przez firmę CAIRE zapośrednictwem dostawcy sprzętu do podawania
tlenu. Korzystanie z akcesoriów niewymienionych
na poniższej liście może mieć negatywny wpływ na
działanie lub bezpieczeństwo koncentratora.

1. Umieścić urządzenie na dole torby, ustawiając jetak, aby otwory na filtry i akumulator znalazły się
we właściwym miejscu.

Standardowe akcesoria FreeStyle Comfort
Torba

MI396-1

PlecakMI459-1

Pasek na ramię

MI397-1

Zasilacz AC (wraz ze wszystkimi
przewodami)

PW036-1S

Zasilacz DC

CD041-1Zewnętrzny filtr dużych cząstek
(prawy)FI226-1Zewnętrzny filtr dużych cząstek (lewy)

FI227-1

2. Zamknąć zamek na dole torby.

Dostępne akcesoria opcjonalne
Dodatkowy akumulator 8-ogniwowy

BT034-1S

Akumulator 16-ogniwowy

BT035-1SŁadowarka biurkowa

BT036-1

Torba na wszystkie akcesoria

MI372-2

Uwaga: Jeśli tak napisano, mogą być dostępne
specjalne przewody zasilania dla poszczególnychkrajów. Jeśli potrzebny jest alternatywny przewód, należy
skontaktować się z firmą CAIRE lub dostawcą sprzętu do
podawania tlenu.

PRZYCZEPIANIE PASKA NA RAMIĘ
1. Pasek na ramię można przyczepić do urządzenia
lub torby.
2. Przypiąć pasek do ucha urządzenia lub torby.

Instrukcje dotyczące torby
Aby umieścić i zabezpieczyć urządzenie FreeStyle
Comfort w torbie, należy postępować zgodnie zponiższymi instrukcjami.

506- PL PN MN232-C4 L | Instrukcja obsługi

FreeStyle Comfort
Utylizacja i ponowne użycie
W celu prawidłowej utylizacji urządzenie FreeStyle
Comfort wraz z wszystkimi komponentami należy
oddać do dostawcy sprzętu. Można też skontaktowaćsię z urzędem gminy lub miasta, aby dowiedzieć się,jak prawidłowo zutylizować akumulator. Jednostkamoże być używana u innych pacjentów, ale musi
zostać zwrócona dostawcy sprzętu w celu regeneracji.

Dyrektywy WEEE i RoHS
Ten symbol przypomina właścicielom sprzętu,że zgodnie z Dyrektywą w sprawie zużytego sprzętu
elektrycznego i elektronicznego (WEEE) po zakończeniu eksploatacji należy go oddać do zakładu utylizacji.
Nasze produkty są zgodne z Dyrektywą w sprawie
ograniczenia stosowania niektórych niebezpiecznych
substancji w sprzęcie elektrycznym i elektronicznym(RoHS). Zawierają jedynie śladowe ilości ołowiu
i innych niebezpiecznych materiałów.

Materiały, z którymi osoba
obsługująca urządzenie ma
bezpośredni lub pośredni kontakt• Obudowa koncentratora: PC/ABS• Panel sterowania koncentratora: Poliester EBG7
lub odpowiednik• Złącze wylotowe tlenu, dysza: Aluminium• Etykieta urządzenia: Lexan• Złącza przewodów: Poliwęglan/polichlorek winylu• Przewód zasilania: PCV, metal• Zasilacz: Lexan 940 (poliwęglan)• Akumulator: PC/ABS• Etykiety na akumulatorze, zasilaczu:
Folia poliestrowa• Walizka transportowa koncentratora: Cordura,
neopren, podszewka nylonowa• Pasek: Polipropylen, Acetyl, winyl, cordura, neopren

PRZESTROGA: Aby dowiedzieć się, jak
prawidłowo zutylizować urządzenie, należy
skontaktować się z dostawcą sprzętu lub
odpowiednim organem władz lokalnych.
PRZESTROGA: Zawsze należy utylizować sito
molekularne zgodnie ze wszystkimi krajowymi,
wojewódzkimi i lokalnymi przepisami dotyczącymi
kontroli środowiska.

PN MN232-C4 L | Instrukcja obsługi PL - 507

FreeStyle Comfort
Testy kompatybilności elektromagnetycznej
Wskazówki i deklaracja producenta — emisja promieniowania elektromagnetycznegoUrządzenie FreeStyle Comfort jest przeznaczone do użytku w środowisku elektromagnetycznym opisanym poniżej.
Klient lub użytkownik urządzenia FreeStyle Comfort powinien upewnić się, że jest ono używane w takim środowisku.
Test emisji

Zgodność

Emisje o częstotliwości
fal radiowych

Grupa 1CISPR 11
Emisje o częstotliwości
fal radiowych
CISPR 11
Emisje harmoniczne
IEC 61000-3-2
Fluktuacja napięcia /emisje migotania
IEC 61000-3-3Środowisko elektromagnetyczne — wskazówkiUrządzenie FreeStyle Comfort wykorzystuje energię o częstotliwości
radiowej wyłącznie do wewnętrznego działania. W związku z
tym emisje o częstotliwości radiowej są niewielkie i nie powinny
powodować żadnych zakłóceń w działaniu znajdującego się w pobliżu
sprzętu elektronicznego.

Klasa B
Klasa A
Zgodne508- PL PN MN232-C4 L | Instrukcja obsługiUrządzenie FreeStyle Comfort jest dostosowane do działania wkażdym środowisku, w tym w warunkach domowych oraz miejscachbezpośrednio podłączanych do publicznej sieci zasilania o niskimnapięciu, która zasila budynki mieszkalne.

FreeStyle Comfort
Wskazówki i deklaracja producenta — odporność elektromagnetyczna

Koncentrator tlenu FreeStyle Comfort jest przeznaczony do użytku w środowisku elektromagnetycznym opisanym poniżej.
Klient lub użytkownik urządzenia FreeStyle Comfort powinien zapewnić używanie go w takim środowisku.
Test odporności
Poziom testu IEC 60601 Poziom zgodności
Środowisko elektromagnetyczne —wskazówki
Podłogi powinny być drewniane, betonoweŚrodowisko±8 kV styk
±8 kV styk
lub wyłożone płytkami ceramicznymi.
elektromagnetyczne
Jeśli podłogi pokryte są materiałem±15 kV przy wylocie±15 kV przy wylocie— wskazówki
syntetycznym, wilgotność względna
powietrza
powietrza
IEC 61000-4-2
powinna wynosić przynajmniej 30%.
±2 kV dla linii zasilających ±2 kV dla linii zasilaNagły skok / impulsJakość zasilania sieciowego powinnającychelektryczny
być typowa dla profesjonalnej placówki±1 kV dla linii wejścia/szpitalnej lub środowiska opieki domowej.
IEC 61000-4-4wyjścia
Nie dotyczy±1 kV linia(-e) do linii

Skok napięcia
IEC 61000-4-5±2 kV tryb wspólny na
liniach AC±1 kV tryb różnicowy na
liniach AC±1 kV linia(-e) do liniiJakość zasilania sieciowego powinna
być typowa dla profesjonalnej placówki
szpitalnej lub środowiska opieki domowej.

±2 kV tryb wspólnyna liniach We/Wy
na zewnątrz
0% UT przez 0,5 cyklu
0% UT przez 0,5 cyklu(0°, 45°, 90°, 135°, 180°, (0°, 45°, 90°, 135°, 180°,
225°, 270°, 315°)
225°, 270°, 315°)Jakość zasilania sieciowego powinna
być typowa dla profesjonalnej placówki
Spadki, krótkie
szpitalnej lub środowiska opieki domowej.
przerwy oraz
0% UT przez 1 cykl (0°)
0% UT przez 1 cykl (0°)
Jeśli użytkownik koncentratora FreeStylezmiany napięcia na
Comfort wymaga ciągłego działania
liniach wejściowych
70% UT (30% zapad UT)
70% UT (30% zapad UT) urządzenia w trakcie przerw w zasilaniu,
zasilania
zaleca się wyposażenie koncentratora
przez 25/30 cykli (0°)
przez 25/30 cykli (0°)
IEC 61000-4-11
FreeStyle Comfort w zasilacz awaryjny(UPS) lub akumulator.
0% UT przez 250/300
0% UT przez 250/300
cykli (0°)
cykli (0°)
Pole magnetyczne o częstotliwości
Pole magnetyczne oźródła zasilania powinno być naczęstotliwości źródła
poziomie charakterystycznym dla typowej
3 A/m
3 A/m
zasilania (50/60 Hz)
profesjonalnej placówki szpitalnej lub
IEC 61000-4-8środowiska opieki domowej.
UWAGA: UT to napięcie sieci AC przed zastosowaniem poziomu testowego.

UWAGA: W niektórych przypadkach urządzenie Freestyle Comfort może zresetować się po dużym wyładowaniu
elektrostatycznym. W celu przywrócenia normalnego działania może być konieczne wykonanie pełnego cyklu zasilania.
Szczegółowe informacje zawarte są w części poświęconej rozwiązywaniu problemów.

PN MN232-C4 L | Instrukcja obsługi PL - 509

FreeStyle Comfort
Wskazówki i deklaracja producenta — odporność elektromagnetyczna

Koncentrator tlenu FreeStyle Comfort jest przeznaczony do użytku w środowisku elektromagnetycznym opisanym poniżej.
Klient lub użytkownik urządzenia FreeStyle Comfort powinien zapewnić używanie go w takim środowisku.
Test odporności Poziom testu IEC 60601 Poziom zgodności
Środowisko elektromagnetyczne —wskazówki
10 Vrms
10 Vrms
Przenośny sprzęt radiokomunikacyjny
Przewodzonaczęstotliwość
nie powinien być używany bliżej dowolnej
od 150 kHz do 80 MHz
radiowaczęści koncentratora tlenu FreeStyle
Comfort, łącznie z przewodami, niżIEC 61000-4-6
zalecana odległość wyliczona ze wzoru
odpowiedniego dla częstotliwości nadajnika.
10 V/m
10 V/mPromieniowanieo częstotliwości
radiowej

Od 80 MHz do 2700 MHz

IEC 61000-4-3

Zalecana odległość pomiędzy urządzeniami
d = 1.2 P

d = 1.2 P = 80 MHz do 800 MHz
d = 1.2 P = 800 MHz do 2700 MHz
gdzie P jest maksymalną wartością mocywyjściowej nadajnika w watach (W)zgodnie z danymi producenta nadajnika,
a d jest zalecaną odległością pomiędzy
urządzeniami w metrach (m).
Natężenia pola pochodzącego zumieszczonych na stałe nadajników
pracujących na częstotliwościach radiowych,określone na podstawie pomiaru poziomu
zakłóceń elektromagnetycznych w miejscu
montażu, a powinny być mniejsze niż poziom
zgodności w każdym zakresie częstotliwości. b
Zakłócenia mogą występować w pobliżu
urządzeń oznaczonych następującymi
symbolami:

UWAGA 1: Przy 80 MHz i 800 MHz stosowany jest zakres większych częstotliwości.
UWAGA 2: Te zalecenia mogą nie mieć zastosowania we wszystkich sytuacjach. Na propagację elektromagnetyczną
wpływa pochłanianie i odbijanie od struktur, przedmiotów i ludzi.
aNatężeń pola generowanych przez nadajniki stałe, takie jak radiostacje bazowe (komórkowe/bezprzewodowe) do
telefonów, przenośne aparaty radiowe, amatorskie stacje radiowe, urządzenia do nadawania na falach średnich i
ultrakrótkich oraz telewizyjnych, nie można teoretycznie przewidzieć w sposób precyzyjny. Aby ocenić środowisko
elektromagnetyczne powstałe na skutek działania nadajników pracujących na falach radiowych, należy rozważyć
przeprowadzenie pomiarów w miejscu instalacji. Jeśli pomiary natężeń pola w miejscu, w którym używany jest koncentrator
serii FreeStyle Comfort, przekraczają odpowiedni poziom zgodności fal radiowych podany powyżej, urządzenie
FreeStyle Comfort należy monitorować pod kątem prawidłowego działania. Jeśli zostanie zaobserwowane nieprawidłowe
działanie, może być konieczne podjęcie dodatkowych środków takich jak zmiana orientacji lub ustawienia urządzenia
FreeStyle Comfort.
b

Przy zakresie częstotliwości od 150 kHz do 80 MHz natężenia pola powinny wynosić mniej niż 10 V/m.

510- PL PN MN232-C4 L | Instrukcja obsługi

FreeStyle Comfort
Zalecane odległości między przenośnym sprzętem komunikacyjnym pracującym naczęstotliwościach radiowych a urządzeniem FreeStyle Comfort

Koncentrator tlenu FreeStyle Comfort powinien być używany w środowisku elektromagnetycznym, w którym zakłócenia
promieniowania o częstotliwościach radiowych są pod kontrolą. Klient lub użytkownik koncentratora tlenu FreeStyle Comfort może pomóc zapobiegać zakłóceniom elektromagnetycznym poprzez utrzymywanie minimalnej odległości pomiędzy
przenośnym sprzętem komunikacyjnym działającym na częstotliwościach radiowych (nadajnikami) a modułem FreeStyle
Comfort w sposób opisany poniżej, zgodnie z maksymalną mocą wyjściową sprzętu komunikacyjnego.
Maksymalna wartość
Odległość w zależności od częstotliwości nadajnika
mocy wyjściowejmnadajnika
od 150 kHz do 80 MHz
od 80 MHz do 800 MHz
od 800 MHz do 2,5 GHz
d = 1.2 P
d = 1.2 P
d = 1.2 P
W
0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
W przypadku nadajników, których maksymalna moc wyjściowa nie jest wymieniona powyżej, zalecaną odległość pomiędzy
urządzeniami w metrach (m) można oszacować, stosując odpowiednie równanie dla częstotliwości nadajnika, gdzie P jest
maksymalną wartością mocy wyjściowej nadajnika w watach (W), zgodnie z danymi producenta nadajnika.
UWAGA 1: Przy 80 MHz i 800 MHz stosowana jest odległość dla zakresu większych częstotliwości.
UWAGA 2: Te zalecenia mogą nie mieć zastosowania we wszystkich sytuacjach. Na propagację elektromagnetyczną
wpływa pochłanianie i odbijanie od struktur, przedmiotów i ludzi.

Klasyfikacja

Typ zabezpieczenia przed porażeniem elektrycznym:
Klasa II	ochrony przed porażeniem prądem
elektrycznym osiągnięta przez podwójną
izolację.
Stopień ochrony przed porażeniem elektrycznym:Sprzęt 	typu BF
zapewniający określony stopień
ochrony przed porażeniem elektrycznym		odnośnie do		1) dopuszczalnego upływu prądu,		2) niezawodności uziemienia ochronnego		(jeżeli jest stosowane).
		Urządzenie nie jest przeznaczone
do bezpośredniego zastosowania
kardiologicznego.
Niezależne testy dla celów sprawdzenia zgodności
z normami dla elektrycznych urządzeń medycznych:	Testy wykazały zgodność z normami• IEC 60601-1 edycja 3.1: Medyczne urządzenia
elektryczne — Cześć 1. Wymagania ogólne
dotyczące bezpieczeństwa.
• CAN/CSA C22.2 No. 60601-1-8 wyd. 2.1,
Medyczne urządzenia elektryczne — Część 1.
Wymagania ogólne dotyczące bezpieczeństwa.

	• Testy wykazały zgodność z RTCA/DO160,
	sekcją 21, kategorią M.
• CISPR 11 / EN 55011, klasa B, grupa 1,
	„Industrial, Scientific, and Medical (ISM) Equipment”• FCC, część 15, punkt B — klasa B.
	Źródło promieniowania niezamierzonego.
IP22 — Ochrona przed wkładaniem palców i innychciał o średnicy większej niż 12,5 mm. Krople wodyspadające pionowo nie mają negatywnego wpływu,gdy obudowa jest odchylona od normalnej pozycji
pod kątem 15°.
Dozwolone metody czyszczenia i zapobiegania
przenoszeniu chorób zakaźnych: Informacjedostępne są w części „Czyszczenie, pielęgnacja
i rutynowe czynności konserwacyjne” niniejszej
instrukcji obsługi urządzenia FreeStyle.
Poziom bezpieczeństwa w przypadku użytkowania
w obecności łatwopalnych gazów znieczulających:	Urządzenie nie jest przeznaczone do zastosowania
w takich warunkach.
Tryb działania: Praca ciągła.

Ochrona przed potencjalnymi zakłóceniami
elektromagnetycznymi i innymi pomiędzy
urządzeniami a innym sprzętem.
	• Testy wykazały zgodność z normą EN 60601-1-2/IEC 60601-1-2: Wyd. 4.
PN MN232-C4 L | Instrukcja obsługi PL - 511

FreeStyle Comfort
Felhasználói kezelőszervek és a rendszer állapotjelzői
ISO 7000
Kezelési útmutató; kezelési utasítások.
Reg. sz.: 1641

ISO 7010 szabvány: Grafikus szimbólumok– Biztonsági színek és biztonsági jelzések –Regisztrált biztonsági jelzések

Gyártási ország és dátum. A „CC”
a gyártási ország kétbetűs országkódját
azonosítja. A gyártás dátumaÉÉÉÉ-HH-NN formátumban
van feltüntetve. Reg. sz.: 6049

Olvassa el a Használati útmutatót.
Reg. sz.: M002
Tartsa távol nyílt lángtól, tűztől,szikrától. A dohányzás és a nyílt lánghasználata tilos! Reg. sz.: P003

Óvja az esőtől, tartsa szárazon.
Reg. sz.: 0626

Az egység közelében vagy annaküzemeltetésekor tilos a dohányzás!
Reg. sz.: P002

Egymásra helyezési korlát darabszám
alapján, ahol az „n” az egymásra
helyezhető egységek maximális számát
jelzi. Reg. sz.: 2403
Katalógusszám. Reg. sz.: 2493

Figyelmeztetés. Reg. sz.: W001
IEC 60417
BF típusú alkalmazott alkatrész(elektromos áramütés elleni védelem
fokozata). Reg. sz.: 5333

Sorozatszám. Reg. sz.: 2498
Tárolási vagy üzemi hőmérséklettartomány. Reg. sz.: 0632
Tárolási páratartalom-tartomány.
Reg. sz.: 2620
Légköri nyomás szerinti korlátozás.
Reg. sz.: 2621
A gyártó neve és címe. Reg. sz.: 3082
Álló helyzetet jelölő nyilak.
Reg. sz.: 0623

II. osztályú berendezés, kettős
szigetelésű. Reg. sz.: 5172
2017/745/EU rendelet
Meghatalmazott képviselő az Európai
Közösségben
Ha a termék egyedi
eszközazonosítójának (UDI)
címkéjén szerepel a CE####
szimbólum, az eszköz megfelel az
orvostechnikai eszközökre vonatkozó
követelményeknek. A CE####
szimbólum a bejelentett szervezet
számát jelöli.
További szimbólumok
Tartsa távol a gyúlékony anyagoktól,
olajtól és zsírtól!

Vigyázat, törékeny! Reg. sz.: 0621

Ne szerelje szét!

Veszélyes anyagokat tartalmaz.
Reg. sz.: 3723

Áramlási sebesség növelése.

Importőr. Reg. sz.: 3725
Áramlási sebesség csökkentése.
O2-kimenet

Menü gomb.
Főkapcsoló. Reg. sz.: 5009
RTCA/
DO-16021. fejezet M
kategóriának
megfelelő

512- HU PN MN232-C4 L | Felhasználói kézikönyv

A POC gyártója megállapította, hogyez az eszköz megfelel az összes
vonatkozó FAA elfogadási kritériumnak
és a repülőgép fedélzetén történő
használatnak.

FreeStyle Comfort
A POC gyártója megállapította, hogyez az eszköz megfelel az összes
vonatkozó FAA elfogadási kritériumnak
és a repülőgép fedélzetén történő
használatnak.

2012/19/EU tanácsi irányelv
WEEE

Ez a szimbólum a berendezés
tulajdonosát arra figyelmezteti,hogy a készüléket annak élettartama
végén az elektromos és elektronikus
készülékek hulladékainak kezeléséről
szóló irányelv (WEEE) szerint egyújrahasznosító létesítménybe kelljuttatnia. A termékeink megfelelnek
a veszélyes anyagok korlátozására
(RoHS) vonatkozó irányelvnek.
Az ólom- és egyéb veszélyesanyagtartalom nem haladja meg a nyomnyi
mennyiséget.

Újrahasznosítási szimbólum
Meghatalmazott képviselő Svájcban(ADD)
Ha az eszközön szerepel a bejelentett
szervezet számát jelző UKCA####számmmal ellátott UKCA jelölés,akkor ez az eszköz megfelel az UKCA
előírásoknak.
21 CFR 801.15
A szövetségi törvény a jelen készülékértékesítését orvos által vagy orvosi
rendelvényre korlátozza.
ETL igazolás
Az amerikai egyesült államokbeli ésa kanadai piacok számára tanúsítva
a vonatkozó amerikai egyesültállamokbeli és kanadai szabványoknak
megfelelően.
FCC-azonosítót tartalmaz: WAP2001
FCC-tájékoztató
A készülék CYBLE-022001-00 félvezetőt
tartalmazhat, a Johnson Technology által gyártott
2450AT18B100 számú antennát is ideértve; megfelelaz FCC szabályzat 15. paragrafusában foglaltaknak.
Az eszköz megfelel a moduláris adókészülékekre
vonatkozó, az FCC DA00-1407 jelű nyilvános
közleményében leírt jóváhagyási követelményeknek.
Az adókészülék az alábbi feltételek teljesülése
esetén működtethető: (1) az eszköz nem okozhat
káros interferenciát, és (2) az eszköznek el kellviselnie minden interferenciát, még a működését
nem kívánatos módon befolyásoló interferenciát is.
A termék egy vagy több USA-beli és nemzetközi
szabadalom oltalma alatt állhat. A vonatkozó
szabadalmak listájának megtekintéséhez, kérjük,
látogasson el weboldalunkra az alábbi címen.
Szabadalmak: www.caireinc.com/corporate/patents

IEC 60601-1-8
Alacsony prioritású technikai riasztás
ISO 15223-1
Orvostechnikai eszköz. Reg. sz.: 5.7.7
Egyedi eszközazonosító szám 5.7.10
A S/NZS 4417.1, AS/NZS 4417.2
ACMA megfelelőségi szabályozási
jelölés - az ACMA előírásoknak való
megfelelést jelzi (ADD)

Vezeték nélküli technológia
Csatlakozás típusa

SSP, iAP2, GATT

Gyakoriság

2402–2480 MHz

Max. RF kimeneti
teljesítmény+4 dBmHatótávolság

10 m (2. osztály)

Ajánlott, hogy a gép legalább 1,1 cm-re (0,43 hüvelyk)
legyen a testtől működés közben. Nem vonatkozik
a maszkokra, a légtömlőre és a tartozékokra.

Berendezéseken használatos grafikus jelképek,IEC 60417
Egyenáram Reg. sz.: 5031
Váltóáram Reg. sz.: 5032
IEC 60601-1
Cseppálló

PN MN232-C4 L | Felhasználói kézikönyv HU - 513

FreeStyle Comfort
A FreeStyle Comfort kicsomagolása
Ellenőrizze, hogy minden felsorolt
alkotóelem megtalálható-e a
csomagolásban. Ha bármelyik
tétel hiányzik, azonnal forduljon aforgalmazóhoz.

Szokásos tartozékok• Hordozható oxigénkoncentrátor• Hordtáska• Vállszíj• 8 cellás akkumulátorcsomag• Váltóáramú tápegység• Váltóáramú tápkábel• Egyenáramú tápegység• Felhasználói kézikönyv

Opcionális tartozékok• Kiegészítő 8 cellás
akkumulátorcsomag• 16 cellás akkumulátorcsomag• Asztali töltő• Tartozék hordtáska

A FreeStyle Comfort bemutatása
Ismerje meg a FreeStyle Comfort legfontosabb funkcióit és a felhasználói felületét.

Elülső nézet
Hordszíj csatlakoztatója

Felhasználói
felületOxigénkimenet

Levegőkimenet

Légbeszívó/szűrők
Légbeszívó/szűrők

Tápcsatlakozó
aljzattakarófedele
Akkumulátor
Akkumulátor
kioldógombja
514- HU PN MN232-C4 L | Felhasználói kézikönyv

FreeStyle ComfortFelső nézet
Kijelző
Áramlási sebesség
növelése

Főkapcsoló

Menü gombÁramlási sebesség
csökkentése

Kijelző akkumulátoros működésnél
Áramlás erőssége

Akkumulátor
töltöttsége

autoSAT jelző
Akkumulátor
töltöttségének
százalékos
kijelzése

Riasztási üzenet

Bólusbejuttatás-jelző

Kijelző külső (váltó- vagy egyenáramú) áramforrásnál

Áramlás erőssége
Akkumulátor
töltöttsége

Riasztási üzenet
helye

Töltésjelző

PN MN232-C4 L | Felhasználói kézikönyv HU - 515

FreeStyle Comfort
Mit jelent az, hogy
oxigénkoncentrátor?
A levegő, amelyet belélegzünk, körülbelül 21%
oxigént, 78% nitrogént és 1% egyéb gázokat tartalmaz.
A FreeStyle® Comfort® esetében a eszköz a szoba
levegőjét szívja be a légbeszívó nyílásokon keresztül.
Ezután ezt egy adszorbens anyagon, úgynevezettmolekuláris szűrőn keresztül vezeti át. Ez az anyag
elkülöníti az oxigént a nitrogéntől, és csak az oxigént
hagyja átáramolni. Ennek eredményeként magas
oxigénkoncentrációjú gázáram jut el a felhasználóhoz.
A FreeStyle Comfort egyesíti a modern
oxigénkoncentrátort és az oxigénmegőrző
technológiákat, hogy egy olyan hordozható készüléket
hozzon létre, amely maximális hordozhatóság
biztosít, hogy a beteg szabadon tudjon mozogni.
A fejlett impulzusáramlású oxigénbeviteli készülék
gyorsan érzékeli a felhasználó belégzését, és minden
belégzés kezdetekor egy oxigénimpulzust adagol.
Az impulzusáramlású gázbevitel ideális a mozgáshoz
és az otthontól távol végzett tevékenységekhez.
A folyamatos gázáramláshoz képest hatékonyabban
gazdálkodik az oxigénnel, és meghosszabbítja az
akkumulátor üzemidejét is. A FreeStyle Comfort
elég könnyű ahhoz, hogy vállszíjon magával vihesse.
Működtethető váltóáramról (fali csatlakozóról),egyenáramról (gépjárműben) vagy akkumulátorról is.

Rendeltetésszerű használat
A CAIRE FreeStyle Comfort oxigénkoncentrátor
kiegészítő oxigén adagolására szolgál. A készülék nem
alkalmas létfenntartó kezelésre, és betegmonitorozó
funkciókkal sem rendelkezik.

Miért írt fel Önnek az orvos oxigént?
Sokan élnek különböző szív-, tüdő- és más légzőszervi
betegséggel. Ezeknek a betegeknek a nagy részénél
hasznos lehet a kiegészítő oxigénterápia otthonukban,
utazás közben vagy az otthonon kívüli napi
tevékenységek során.
Az oxigén egy gáz, amely az általunk belélegzett
levegő 21%-át teszi ki, és amely folyamatosanszükséges ahhoz, hogy a szervezetünk megfelelően
tudjon működni. Orvosa kifejezetten az Ön légzőszervi
problémájának megfelelő áramlási sebességet vagybeállítást írt fel Önnek.

516- HU PN MN232-C4 L | Felhasználói kézikönyv

Bár az oxigén használata nem okoz függőséget,a felügyelet nélkül végzett oxigénterápia veszélyes
lehet. Az oxigénkoncentrátor használata előtt
forduljon orvoshoz és kérjen tanácsot. A berendezés
forgalmazója, aki leszállítja az oxigénberendezést,
megmutatja Önnek, hogy hogyan kell a felírt áramlási
sebességet beállítani.

Kezelői profil
A koncentrátor arra szolgál, hogy kiegészítő oxigént
juttasson azoknak a betegeknek, akiknél valamilyen
betegség károsan befolyásolja tüdejük arra irányuló
képességét, hogy a levegő oxigénjét a vérbe
juttassa, és ezért panaszaik vannak. A hordozható
oxigénkoncentrátorok (POC készülékek) nem tárolnak
és nem tartalmaznak oxigént. Nem kell utántölteni
ezeket, és az akkumulátoruk bármikor feltölthető az
elérhető váltó- vagy egyenáramú áramforrásról (kivévea repülőgépeken). Ezáltal a felhasználó viszonylagosönellátásra képes akár otthoni, akár mozgás közbeni
használat esetén (mind a házon belül, mind azon
kívül), mobilissé válik, és általánosságban elmondható,
hogy a megszokott életét élheti. Az oxigénkoncentrátor
használatához orvosi rendelvény szükséges.
Az oxigénterápia felhasználók általában 65 évnél
idősebb betegek, akiknek különféle légzőrendszeri
betegségük van, mint például a krónikus obstruktívtüdőbetegség (COPD). A felhasználók általában
jó kognitív képességgel rendelkeznek, és képesnekkell lenniük jelezni, ha rosszul érzik magukat. Ha abeteg nem képes közölni, ha rosszul van, vagy ha
nem tudja elolvasni és megérteni a koncentrátor
címkéjét és kezelési útmutatóját, akkor a használatcsak olyan személy felügyelete mellett javasolt, aki
képes ezekre. Ha a koncentrátor használata közben
bármilyen kellemetlenséget érez, akkor javasoljuk,
hogy beszéljen egészségügy szakemberrel. Szintén
javasoljuk, hogy rendelkezzen tartalék oxigénforrással
(azaz oxigénpalackkal) arra az esetre, ha áramszünet
van, vagy a koncentrátor meghibásodik. A koncentrátor
használatához nincs szükség semmilyen egyéb
különleges készségre vagy képességre.

FreeStyle Comfort
Fontos!
A következő biztonsági figyelmeztetések vannakérvényben:
FIGYELMEZTETÉS: FONTOS BIZTONSÁGI
INFORMÁCIÓK OLYAN VESZÉLYEKKELKAPCSOLATBAN, AMELYEK SÚLYOSEGÉSZSÉGKÁROSODÁST OKOZHATNAK.
VIGYÁZAT: Fontos tudnivalók, hogy
elkerülhesse a FreeStyle Comfort
megrongálódását.
Megjegyzés: Különös figyelmet igénylő információk.

Alkalmazási terület
FIGYELMEZTETÉS: NAGYON FONTOS,
HOGY CSAK A FELÍRT OXIGÉNSZINTETÁLLÍTSA BE. CSAK AKKOR MÓDOSÍTSA A
KIVÁLASZTOTT ÁRAMLÁSI SEBESSÉGET,HA EGY ILLETÉKES ORVOSTÓL ERREKAP UTASÍTÁST.
FIGYELMEZTETÉS: EZT AZ ESZKÖZT NEM TESZTELTÉKVAGY HAGYTÁK JÓVÁ GYERMEKGYÓGYÁSZATIBETEGEKNÉL VALÓ HASZNÁLATRA.
FIGYELMEZTETÉS: AZ AMERIKAI EGYESÜLT
ÁLLAMOK SZÖVETSÉGI TÖRVÉNYEI A JELEN
BERENDEZÉS FORGALMAZÁSÁT ÉS BÉRLETÉTCSAK ORVOS VAGY MÁS EGÉSZSÉGÜGYI
SZAKEMBER RENDELVÉNYÉRE ENGEDÉLYEZIK.
FIGYELMEZTETÉS: A KÉSZÜLÉKET NEM SZABAD
ÉLETFENNTARTÁSRA HASZNÁLNI. HA AZESZKÖZT OLYAN FELHASZNÁLÓ HASZNÁLJA,AKI NEM KÉPES JELEZNI, HA ROSSZUL VAN AZESZKÖZ HASZNÁLATA KÖZBEN, AKKOR TOVÁBBI
MONITOROZÁSRA LEHET SZÜKSÉG. A HALLÁS- ÉS/
VAGY LÁTÁSSÉRÜLT FELHASZNÁLÓK SEGÍTSÉGRESZORULHATNAK A RIASZTÁSOK FIGYELÉSÉBEN.
FIGYELMEZTETÉS: MINDEN FELHASZNÁLÓESETÉBEN EGYÉNILEG KELL MEGHATÁROZNIAZ IMPULZUS-MÓD BEÁLLÍTÁSAIT A NYUGALMI,FIZIKAI AKTIVITÁS ÉS UTAZÁS KÖZBENISZÜKSÉGLETÜKNEK MEGFELELŐEN.

FIGYELMEZTETÉS: A BETEG EGYES LÉGZÉSIERŐFESZÍTÉSEI ESETLEG NEM INDÍTJÁK EL
AZ OXIGÉNMEGTAKARÍTÁSI FUNKCIÓT.
FIGYELMEZTETÉS: A FREESTYLE COMFORT NEM
HASZNÁLHATÓ TRACHEOTOMIZÁLT BETEGEKNÉL.
FIGYELMEZTETÉS: TERHES VAGY SZOPTATÓ NŐK
NEM HASZNÁLHATJÁK A JELEN KÉZIKÖNYVBENAJÁNLOTT TARTOZÉKOKAT, MERT AZOKFTALÁTOKAT TARTALMAZHATNAK.

A használat ellenjavallatai
FIGYELMEZTETÉS: A NEM ORVOSÁLTAL ELRENDELT OXIGÉNHASZNÁLAT
BIZONYOS KÖRÜLMÉNYEK KÖZÖTT
VESZÉLYES IS LEHET. A KÉSZÜLÉKETCSAK ORVOS ÁLTAL ELRENDELTESETBEN SZABAD HASZNÁLNI.
FIGYELMEZTETÉS: EZ AZ ESZKÖZ NEM
HASZNÁLHATÓ MÁGNESES REZONANCIÁSKÖRNYEZETBEN VAGY ANNAK KÖZELÉBEN,PÉLDÁUL MR KÉSZÜLÉK MELLETT.
FIGYELMEZTETÉS: NE HASZNÁLJA A KÉSZÜLÉKETGYÚLÉKONY ALTATÓSZEREK JELENLÉTÉBEN!
FIGYELMEZTETÉS: MINT MINDEN ELEKTROMOS KÉSZÜLÉK ESETÉBEN, A FELHASZNÁLÓ ITT IS TAPASZTALHAT OLYAN IDŐSZAKOKAT, AMIKOR A KÉSZÜLÉK
NEM MŰKÖDIK. ENNEK LEHET OKA ÁRAMSZÜNET,ILLETVE SZERVIZMUNKÁRA IS SZÜKSÉG LEHET,AMELYET SZAKKÉPZETT TECHNIKUSNAK KELL
ELVÉGEZNIE. AZ OXIGÉNKONCENTRÁTOR HASZNÁLATA NEM MEGFELELŐ OLYAN FELHASZNÁLÓ
SZÁMÁRA, AKINÉL AZ ILYEN ÁTMENETI ÜZEMKIESÉSEK SÚLYOS EGÉSZSÉGÜGYI KÖVETKEZMÉNYEKKEL
JÁRNAK.
FIGYELMEZTETÉS: EZ AZ ESZKÖZ NEM MINDENKÖRNYEZETBEN HASZNÁLHATÓ. BIZONYOSELEKTROMÁGNESES FORRÁSOKKAL VALÓÉRINTKEZÉS INTERFERENCIÁT OKOZHAT VAGYAZ ESZKÖZ NEM KÍVÁNATOS TELJESÍTMÉNYÉTEREDMÉNYEZHETI. INTERFERENCIA ESETÉNAZ ESZKÖZT AZONNAL KI KELL KAPCSOLNI.

FIGYELMEZTETÉS: EZT AZ ESZKÖZT OTTHONIÁPOLÁSI ÉS HASONLÓ JÁRÓBETEG-ELLÁTÁSIKÖRNYEZETBEN HASZNÁLATRA ALAKÍTOTTÁK KI.
FIGYELMEZTETÉS: EZT AZ ESZKÖZT AZOXIGÉNTERÁPIÁBAN RÉSZESÜLŐ BETEGNEK KELL
HASZNÁLNIA.

PN MN232-C4 L | Felhasználói kézikönyv HU - 517

FreeStyle Comfort
Biztonsági irányelvek
FIGYELMEZTETÉS: A KÉSZÜLÉKEN TILOS
MÓDOSÍTÁST VÉGEZNI!
FIGYELMEZTETÉS: EZ A KÉSZÜLÉK NAGY
KONCENTRÁCIÓJÚ OXIGÉNT AD LE, AMI ELŐSEGÍTIA GYORS ÉGÉST. NE ENGEDJE MEG A DOHÁNYZÁST
VAGY NYÍLT LÁNG HASZNÁLATÁT ABBAN AHELYISÉGBEN, AHOL (1) EZ A KÉSZÜLÉK VAGY (2)BÁRMILYEN OXIGÉNT SZÁLLÍTÓ TARTOZÉK VAN.
ENNEK A FIGYELMEZTETÉSNEK A FIGYELMEN KÍVÜL
HAGYÁSA SÚLYOS TÜZET, ANYAGI KÁROKAT ÉS/VAGY FIZIKAI SÉRÜLÉST VAGY HALÁLT OKOZHAT.
FIGYELMEZTETÉS: NE MŰKÖDTESSE AZ
EGYSÉGET KISMÉRETŰ VAGY ZÁRT TEREKBEN(PL. KIS TOKBAN VAGY KÉZITÁSKÁBAN), AHOL
KORLÁTOZOTT A LÉGCSERE LEHETŐSÉGE.
EZ UGYANIS AZ OXIGÉNKONCENTRÁTOR
TÚLHEVÜLÉSÉT ÉS A TELJESÍTMÉNYÉNEKROMLÁSÁT OKOZHATJA.
FIGYELMEZTETÉS: A KONCENTRÁTORT OLYANHELYRE KELL TENNI, AHOL EL LEHET KERÜLNIA FÜSTÖT, A SZENNYEZŐ ANYAGOKAT ÉS A PÁRÁT.
FIGYELMEZTETÉS: ABBAN AZ ESETBEN, HA
SÚLYOS ESEMÉNY TÖRTÉNIK A KÉSZÜLÉKKEL,A FELHASZNÁLÓNAK AZONNAL JELENTENIEKELL AZ ESETET A SZOLGÁLTATÓNAK ÉS/VAGY A GYÁRTÓNAK. A SÚLYOS ESEMÉNYFOGALMA: SÉRÜLÉS, HALÁL VAGY SÉRÜLÉS/HALÁL OKOZÁSÁNAK LEHETŐSÉGE, HA AZESEMÉNY ÚJRA ELŐFORDUL. A FELHASZNÁLÓ AZ
ESEMÉNYT BEJELENTHETI ANNAK AZ ORSZÁGNAK
AZ ILLETÉKES HATÓSÁGÁNÁL IS, AHOL AZ
ESEMÉNY TÖRTÉNT.
FIGYELMEZTETÉS: A NEM EHHEZAZ OXIGÉNKONCENTRÁTORHOZ VALÓEGYES OXIGÉNBEVITELI TARTOZÉKOKHASZNÁLATA BEFOLYÁSOLHATJAA TELJESÍTMÉNYT. A JAVASOLTTARTOZÉKOK LISTÁJA MEGTALÁLHATÓEBBEN A KÉZIKÖNYVBEN.
FIGYELMEZTETÉS: HA AZ OXIGÉNKONCENTRÁTORT
LEEJTETTÉK, MEGSÉRÜLT VAGY VÍZ ÉRTE,FORDULJON AZ ESZKÖZ FORGALMAZÓJÁHOZ
ELLENŐRZÉS ÉS AZ ESETLEGESJAVÍTÁS ÉRDEKÉBEN. NE HASZNÁLJA AZOXIGÉNKONCENTRÁTORT, HA A TÁPKÁBEL
VAGY A CSATLAKOZÓJA MEGSÉRÜLT.
FIGYELMEZTETÉS: AZ ÖVVEL VAGY EGYÉBMÓDON MEGFELELŐEN RÖGZÍTSE AZOXIGÉNKONCENTRÁTORT, HA GÉPJÁRMŰBENHASZNÁLJA, HOGY ELKERÜLJE A SÉRÜLÉST VAGYA KÉSZÜLÉK KÁROSODÁSÁT.

518- HU PN MN232-C4 L | Felhasználói kézikönyv

FIGYELMEZTETÉS: AZ AMERIKAI EGYESÜLT
ÁLLAMOK KÖZLEKEDÉSI MINISZTÉRIUMÁNAK
(DOT) ÉS AZ EGYESÜLT NEMZETEK (UN) ELŐÍRÁSAIMEGKÖVETELIK AZ AKKUMULÁTOR KIVÉTELÉT A
KÉSZÜLÉKBŐL REPÜLŐN TÖRTÉNŐ UTAZÁSKOR,HA AZ OXIGÉNKONCENTRÁTORT CSOMAGKÉNTADJÁK FEL. AZ OXIGÉNKONCENTRÁTOR
SZÁLLÍTÁSAKOR IS KI KELL VENNI AZ
AKKUMULÁTORT A KÉSZÜLÉKBŐL, ÉS
MEGFELELŐEN BE KELL CSOMAGOLNI.
FIGYELMEZTETÉS: NE HASZNÁLJA A BERENDEZÉSTADDIG, AMÍG EL NEM OLVASTA ÉS MEG NEM
ÉRTETTE EZT A KÉZIKÖNYVET. HA NEM ÉRTI A
FIGYELMEZTETÉSEKET ÉS AZ UTASÍTÁSOKAT,FORDULJON A BERENDEZÉS FORGALMAZÓJÁHOZ,MIELŐTT MEGKÍSÉRELNÉ HASZNÁLNI; ELLENKEZŐESETBEN EGÉSZSÉGKÁROSODÁS VAGY A
BERENDEZÉS KÁROSODÁSA KÖVETKEZHET BE.
FIGYELMEZTETÉS: HA ROSSZUL ÉRZI MAGÁT VAGYEGÉSZSÉGÜGYI VÉSZHELYZETBE KERÜL, AZONNALHÍVJON ORVOSI SEGÍTSÉGET.
FIGYELMEZTETÉS: HA A MŰSZAKI JELLEMZŐKNÉLMEGADOTT ÜZEMELTETÉSI HŐMÉRSÉKLETENKÍVÜL MŰKÖDTETI AZ OXIGÉNKONCENTRÁTORT,AKKOR A KONCENTRÁTOR CSAK KORLÁTOZOTTANTUD MEGFELELNI AZ OXIGÉNKONCENTRÁCIÓSJELLEMZŐKNEK. A HŐMÉRSÉKLETIHATÁRÉRTÉKEKET MEGTALÁLJA A KÉZIKÖNYVMŰSZAKI JELLEMZŐKRŐL SZÓLÓ RÉSZÉBEN.
FIGYELMEZTETÉS: NE HASZNÁLJON OLAJAT, ZSÍRT,PETRÓLEUMALAPÚ VAGY MÁS ÉGHETŐ TERMÉKETAZ OXIGÉNT SZÁLLÍTÓ TARTOZÉKOKHOZ VAGY AZOXIGÉNKONCENTRÁTORHOZ. CSAK OXIGÉNNELÖSSZEFÉRHETŐ, VÍZALAPÚ KRÉMEKET ÉSKENŐCSÖKET SZABAD HASZNÁLNI. AZ OXIGÉNFELGYORSÍTJA AZ ÉGHETŐ ANYAGOK ÉGÉSÉT.
FIGYELMEZTETÉS: AZ AKKUMULÁTOR HELYTELEN
HASZNÁLATA MIATT FELHEVÜLHET VAGYMEGGYULLADHAT AZ AKKUMULÁTOR, ÉS SÚLYOS
SÉRÜLÉST OKOZHAT. NE LYUKASSZA KI, NE
ÜSSÖN RÁ, NE TAPOSSON RÁ, NE EJTSE LE AZAKKUMULÁTORT, ÉS MÁS ERŐBEHATÁSNAK VAGYRÁZKÓDÁSNAK SE TEGYE AZT KI. A KÁROSODOTTAKKUMULÁTOR HASZNÁLATA SZEMÉLYI SÉRÜLÉST
OKOZHAT.
FIGYELMEZTETÉS: FÉMTÁRGGYAL, PL. KULCCSALVAGY PÉNZÉRMÉVEL NE ZÁRJA RÖVIDRE AZ
AKKUMULÁTOR ÉRINTKEZŐIT. EZ SZIKRÁZÁST
OKOZHAT VAGY HŐ TERMELŐDÉSÉHEZ VEZETHET.

FreeStyle Comfort
FIGYELMEZTETÉS: HA A HORDOZHATÓOXIGÉNKONCENTRÁTORT VÁLTÓÁRAMRÓLHASZNÁLJA A HÁZON KÍVÜL, AKKOR A TÁPEGYSÉGETCSAK RÖVIDZÁRLATI MEGSZAKÍTÓVAL ELLÁTOTTCSATLAKOZÓBA CSATLAKOZTASSA.
FIGYELMEZTETÉS: AZ OXIGÉNKONCENTRÁTOR
NEM HASZNÁLHATÓ MÁS BERENDEZÉS MELLETT,ALATT VAGY FÖLÖTT ELHELYEZVE. HA NEM
LEHET ELKERÜLNI, HOGY A KÉSZÜLÉK MÁS
BERENDEZÉS MELLETT, ALATT VAGY FELETTLEGYEN ELHELYEZVE, ELLENŐRIZNI KELL, HOGY
MEGFELELŐEN MŰKÖDIK-E.
FIGYELMEZTETÉS: A MEGADOTTAKTÓL ELTÉRŐ
KÁBELEK ÉS ADAPTEREK ALKALMAZÁSA – A
GYÓGYÁSZATI VILLAMOS KÉSZÜLÉK GYÁRTÓJA
ÁLTAL A BELSŐ ALKATRÉSZEK CSERÉJÉREÁRUSÍTOTT KÁBELEK ÉS ADAPTEREKKIVÉTELÉVEL – AZ OXIGÉNKONCENTRÁTORMEGNÖVEKEDETT ZAVARKIBOCSÁTÁSÁT ÉSCSÖKKENT ZAVARTŰRÉSÉT EREDMÉNYEZHETI.
FIGYELMEZTETÉS: CSAK A KÉSZÜLÉKREHELYEZETT, MŰSZAKI ADATOKAT TARTALMAZÓ
CÍMKÉN MEGHATÁROZOTT ELEKTROMOS
FESZÜLTSÉGET SZABAD HASZNÁLNI.
FIGYELMEZTETÉS: NE HASZNÁLJON AZ EGYSÉGHEZ
HOSSZABBÍTÓT, ÉS NE CSATLAKOZTASSON TÚL
SOK DUGÓT UGYANAHHOZ AZ ELEKTROMOSALJZATHOZ. A HOSSZABBÍTÓ HASZNÁLATA
KÁROSAN BEFOLYÁSOLHATJA A KÉSZÜLÉK
TELJESÍTMÉNYÉT. HA TÚL SOK FOGYASZTÓ
VAN CSATLAKOZTATVA EGY ALJZATHOZ, AZ
AZ ELEKTROMOS PANEL TÚLTERHELÉSÉTEREDMÉNYEZHETI, AMITŐL A MEGSZAKÍTÓ/
BIZTOSÍTÉK AKTIVÁLÓDHAT, ILLETVE TŰZ
KELETKEZHET, HA A MEGSZAKÍTÓ VAGY A
BIZTOSÍTÉK MEGHIBÁSODIK.
VIGYÁZAT: Ha riasztási esemény
fordul elő, vagy ha Ön azt észleli, hogyaz oxigénkoncentrátor nem működik
megfelelően, olvassa el a kézikönyv
Hibaelhárítás című fejezetét. Ha nem
tudja megoldani a problémát, forduljon
a berendezés forgalmazójához.
VIGYÁZAT: A garancia érvényességének megőrzéseérdekében kövesse a gyártó minden utasítását.

FIGYELMEZTETÉS: NE KÍSÉRELJEN MEG
ELVÉGEZNI SEMMILYEN KARBANTARTÁSI
MUNKÁLATOT A JELEN KÉZIKÖNYVBENFELSOROLTAKON KÍVÜL.
FIGYELMEZTETÉS: CSAK A GYÁRTÓ ÁLTAL
AJÁNLOTT TARTOZÉKOK. AZ EGYÉB TARTOZÉKOKHASZNÁLATA VESZÉLYES LEHET, KÁROSODHATAZ OXIGÉNKONCENTRÁTOR, ÉS A GARANCIAELVESZTÉSÉVEL JÁR.
FIGYELMEZTETÉS: A GYÁRTÓ AZT JAVASOLJA,HOGY TARTSON MAGÁNÁL KIEGÉSZÍTŐOXIGÉNFORRÁST ARRA AZ ESETRE, HAÁRAMSZÜNET LENNE, ILLETVE RIASZTÁSI
KÖRÜLMÉNY VAGY MŰSZAKI HIBA LÉPNE FEL.
ÉRDEKLŐDJÖN ORVOSÁNÁL VAGY A BERENDEZÉSFORGALMAZÓJÁNÁL, HOGY MILYEN TÍPUSÚTARTALÉKRENDSZERRE VAN SZÜKSÉGE.
FIGYELMEZTETÉS: NE HAGYJA, HOGY AKÁRA LEVEGŐBEMENETI, AKÁR A LÉGKIMENETINYÍLÁSOK TAKARVA LEGYENEK. NE EJTSENVAGY HELYEZZEN BE SEMMILYEN TÁRGYAT AKÉSZÜLÉK SEMELYIK NYÍLÁSÁBA. EZ UGYANISAZ OXIGÉNKONCENTRÁTOR TÚLHEVÜLÉSÉT ÉS ATELJESÍTMÉNYÉNEK ROMLÁSÁT OKOZHATJA.
FIGYELMEZTETÉS: AZ OXIGÉN HASZNÁLATAKÖZBENI DOHÁNYZÁS A TŰZZEL KAPCSOLATOSSÉRÜLÉSEKNEK ÉS HALÁLESETEKNEK AZ ELSŐSZÁMÚ KIVÁLTÓ OKA. AZ ALÁBBI BIZTONSÁGI
FIGYELMEZTETÉSEKET SZIGORÚAN BE KELLTARTANI:FIGYELMEZTETÉS: NE ENGEDJE MEG ADOHÁNYZÁST, GYERTYA VAGY NYÍLT LÁNGHASZNÁLATÁT ABBAN A HELYISÉGBEN, AHOLA KÉSZÜLÉKET VAGY OXIGÉNT SZÁLLÍTÓTARTOZÉKAIT HASZNÁLJA.
FIGYELMEZTETÉS: A DOHÁNYZÁS AZ OXIGÉNKANÜL
VISELÉSE KÖZBEN AZ ARC MEGÉGÉSÉTOKOZHATJA ÉS HALÁLHOZ VEZETHET.
FIGYELMEZTETÉS: HA DOHÁNYZIK, AZ ALÁBBI
3 LÉPÉS ÉLETMENTŐ LEHET: KAPCSOLJA KI AZOXIGÉNKONCENTRÁTORT, VEGYE LE A KANÜLT,ÉS HAGYJA EL A HELYISÉGET, AHOL A KÉSZÜLÉK
TALÁLHATÓ.

PN MN232-C4 L | Felhasználói kézikönyv HU - 519

FreeStyle Comfort
FIGYELMEZTETÉS: „TILOS A DOHÁNYZÁS –OXIGÉNHASZNÁLAT” FELIRATÚ TÁBLÁKATKELL JÓL LÁTHATÓ MÓDON ELHELYEZNI AFELHASZNÁLÓ OTTHONÁBAN, VAGY AHOL AZOXIGÉNT HASZNÁLJA. A FELHASZNÁLÓKATÉS GONDOZÓIKAT TÁJÉKOZTATNI KELL, HOGYMILYEN VESZÉLYEKKEL JÁR AZ ORVOSI OXIGÉNJELENLÉTÉBEN VAGY ANNAK HASZNÁLATAKÖZBEN TÖRTÉNŐ DOHÁNYZÁS.
FIGYELMEZTETÉS: NE HASZNÁLJA AZOXIGÉNKONCENTRÁTORT GYÚLÉKONY GÁZOKJELENLÉTÉBEN. ELLENKEZŐ ESETBEN GYORSÉGÉS ALAKULHAT KI, AMI ANYAGI KÁROKAT,SZEMÉLYI SÉRÜLÉST VAGY HALÁLT IS OKOZHAT.
FIGYELMEZTETÉS: A KANÜL LEVÉTELE ÉSRUHÁRA, ÁGYNEMŰRE, KANAPÉRA VAGY MÁSPÁRNAANYAGRA HELYEZÉSE FELLOBBANÓ TÜZET
OKOZ, HA CIGARETTÁVAL, HŐFORRÁSSAL VAGYLÁNGGAL KERÜL ÉRINTKEZÉSBE.
FIGYELMEZTETÉS: NE HAGYJA AZ ORRKANÜLT ARUHÁZATON, ÁGYTAKARÓN VAGY SZÉKHUZATON,
ILLETVE AZOK ALATT. HA AZ EGYSÉGETBEKAPCSOLJA, DE NEM HASZNÁLJA, AKKOR AZOXIGÉN EZEKET AZ ANYAGOKAT GYÚLÉKONNYÁTESZI. ELLENŐRIZZE, HOGY KIKAPCSOLTA-E AZOXIGÉNKONCENTRÁTORT, HA NEM HASZNÁLJA.
FIGYELMEZTETÉS: A JÓVÁHAGYOTTOXIGÉNADAGOLÓ TARTOZÉKOKAT ÚGYALAKÍTOTTÁK KI, HOGY A BETEG BIZTONSÁGAÉRDEKÉBEN CSÖKKENTSÉK A TŰZ TERJEDÉSÉT.
A NEM JÓVÁHAGYOTT TARTOZÉKOK HASZNÁLATANÖVELHETI A TŰZHÖZ KAPCSOLÓDÓ SÉRÜLÉSEKVAGY KÁROK KOCKÁZATÁT.
FIGYELMEZTETÉS: AZ AKKUMULÁTORCSOMAG
ELTÁVOLÍTÁSA ELŐTT AZ OXIGÉNKONCENTRÁTORT
KI KELL KAPCSOLNI ÉS LE KELL VÁLASZTANI A FALICSATLAKOZÓRÓL.
FIGYELMEZTETÉS: A KIS ALKATRÉSZEKFULLADÁSVESZÉLYT OKOZHATNAK.
FIGYELMEZTETÉS: NE ÉRINTSE MEG A LÉGKIMENETINYÍLÁST 10 MÁSODPERCNÉL HOSSZABB IDEIG,
MERT MELEG LEHET.
FIGYELMEZTETÉS: A SZÉL VAGY AZ ERŐS HUZATKÁROSAN BEFOLYÁSOLHATJA A PONTOSOXIGÉNSZÁLLÍTÁST AZ OXIGÉNTERÁPIA SORÁN.
FIGYELMEZTETÉS: EBBŐL A TERMÉKBŐL KÉMIAIANYAGOK – TÖBBEK KÖZÖTT NIKKEL – OLDÓDHATNAK KI, AMI KALIFORNIA ÁLLAMBAN ISMERTENRÁKKELTŐ. TOVÁBBI INFORMÁCIÓÉRT LÁTOGASSON EL A WWW.P65WARNINGS.CA.GOV OLDALRA.
VIGYÁZAT: Először mindig válassza lea tápegységet a fali csatlakozóról, éscsak utána válassza le a tápegységet az
oxigénkoncentrátorról.

520- HU PN MN232-C4 L | Felhasználói kézikönyvVIGYÁZAT: Ne állítsa az oxigénkoncentrátort olyan
pozícióba, ahol nehezen hozzáférhető a tápkábel.
VIGYÁZAT: Az oxigénszállító csöveket és a tápkábelt
mindig úgy helyezze el, hogy ne lehessen bennük
megbotlani, és ne tudjanak véletlenül rátekerednia beteg nyakára.
VIGYÁZAT: Ha autóban, hajóban vagy más
egyenáramú áramforrásról használja az
oxigénkoncentrátort, akkor gondoskodjon arról, hogyaz oxigénkoncentrátor bekapcsolása előtt beindítjaés járatja a járművet. Ha nem tartja be ezt az utasítást,
akkor előfordulhat, hogy az áramforrás nem biztosít
energiát az oxigénkoncentrátornak.
VIGYÁZAT: Ha leállítja a járművet, amelyben az
oxigénkoncentrátort használja, akkor válassza
le az eszközt és vegye ki a járműből. Ne tartsa az
oxigénkoncentrátort nagyon meleg vagy nagyon
hideg járműben, vagy más magas vagy alacsonyhőmérsékletű környezetben.
Megjegyzés: A hordozható és mobil RF távközlési
berendezések befolyásolhatják a gyógyászati villamos
készülékek működését.
VIGYÁZAT: NE hagyja az oxigénkoncentrátort
vagy a tápegységet a járműre csatlakoztatva,
ha a gyújtás KI van kapcsolva. Ebben az
esetben lemerülhet a jármű akkumulátora.
VIGYÁZAT: Az oxigénkoncentrátornak a normálhőmérsékleti és környezeti nyomás (tengerszint feletti
magasság) tartományon kívül történő működtetése
vagy tárolása hatással lehet a teljesítményre és
csökkentheti az akkumulátor üzemidejét, és/vagymegnőhet az akkumulátor feltöltési ideje. A tárolási ésüzemeltetési hőmérsékleti határértékeket megtalálja
a kézikönyv Műszaki jellemzők című részében.
VIGYÁZAT: Az akkumulátorral felszerelt
oxigénkoncentrátorok esetében: Hűvös és száraz
helyen tárolja, hogy biztosítsa az akkumulátorélettartamát. Ha hosszabb ideig meleg helyen tárolja
az oxigénkoncentrátort, vagy teljesen feltöltött,illetve teljesen lemerült akkumulátorral tárolja, az
csökkentheti az akkumulátor élettartamát. Ne próbálja
meg felnyitni az akkumulátort; nincsenek bennejavítható alkatrészek. Az akkumulátort tartsa távola gyermekektől.
VIGYÁZAT: CSAK a gyártó által biztosított
akkumulátort HASZNÁLJA. Az akkumulátor helyesártalmatlanításához vegye fel a kapcsolatot aberendezés forgalmazójával, vagy a veszélyes
hulladékok ügyében illetékes helyi állami hivatallal.

FreeStyle Comfort
Megjegyzés: A koncentrátor vagy az áramkimaradásáltal okozott károk enyhítése érdekében a berendezés
forgalmazójának fel kell mérnie a tartalék oxigén
szükségességét a telepítés során, majd rendszeresenaz alábbi jellemzők megváltozása esetén:

Megjegyzés: Ha az oxigénkoncentrátort hosszú időnát a műszaki adatok táblázatában látható normálisüzemeltetési körülményeken kívül eső feltételek között
tárolták, akkor bekapcsolás előtt meg kell várni, hogy
elérje a normális üzemeltetési hőmérsékletet.

- A beteg állapota

Megjegyzés: A berendezés forgalmazójának fel kellmérnie a tartalék oxigén szükségességét a telepítéssorán, majd rendszeresen az alábbi tulajdonságok
megváltozása esetén:- A környezet, amelyben a beteg él- Az oxigénellátás mellett tartalék oxigén is biztosíthatóa betegnek.
Megjegyzés: Ellenőrizze, hogy a kanült teljesenbehelyezte-e és megfelelően rögzül-e. Ez biztosítja,hogy az oxigénkoncentrátor megfelelően tudja érzékelni
a belégzést az oxigénellátáshoz. Belégzés alatt hallaniaés éreznie kell az oxigén áramlását az orrkanül ágaiban.

- A beteg állapota- A környezet, amelyben a beteg él- Az oxigénellátás mellett tartalék oxigén is biztosíthatóa betegnek.

Megjegyzés: Ha az akkumulátor teljesen fel van töltve
vagy teljesen lemerült, és szélsőségesen meleghőmérséklet éri, akkor a lítium-akkumulátorok kapacitása
maradandóan csökkenhet. Hosszú idejű tárolás esetén
az akkumulátorok 25–50%-os feltöltöttsége és a
-20 °C – 20 °C (-4 °F – 68 °F) hőmérsékleti tartomány
az ajánlott.
Megjegyzés: Ne működtesse az oxigénkoncentrátort
behelyezett légbeszívó szűrő nélkül! Ha van másikszűrője is, akkor a szennyezett szűrő megtisztítása
előtt helyezze be a pótszűrőt. Meleg, szappanos vízzel
tisztítsa meg a szennyezett szűrőt, majd használat előtt
teljesen szárítsa meg.
Megjegyzés: A gyártó nem javasolja a berendezéssterilizálását.
Megjegyzés: Attól soha nem kell tartani, hogy az
oxigénkoncentrátor használata közben a helyiségben
elfogy az oxigén.
FIGYELMEZTETÉS: AZOXIGÉNKONCENTRÁTOR NEM SZABAD
MŰKÖDTETNI, VAGY KÜLSŐ (VÁLTÓVAGY EGYENÁRAMÚ) ÁRAMFORRÁSRACSATLAKOZTATNI, HA AZ AKKUMULÁTOR
NINCS BEHELYEZVE. NE PRÓBÁLJA
MEG BEKAPCSOLNI VAGY KÜLSŐ
ÁRAMFORRÁSRA CSATLAKOZTATNIAZ EGYSÉGET, HA KIVETTE BELŐLEAZ AKKUMULÁTORCSOMAGOT.
CSAK AKKOR VEGYE KI AZAKKUMULÁTORCSOMAGOT, HA KIKAPCSOLTAAZ EGYSÉGET, ÉS KIHÚZTA AZT AZ ELEKTROMOSALJZATBÓL. NE PRÓBÁLJA MEG ÚGY KIVENNI AZ
AKKUMULÁTORCSOMAGOT, HOGY AZ EGYSÉGMÉG BE VAN KAPCSOLVA ÉS NINCS KIHÚZVA AZ
ELEKTROMOS ALJZATBÓL.

PN MN232-C4 L | Felhasználói kézikönyv HU - 521

FreeStyle ComfortMűszaki jellemzők
FreeStyle Comfort
Méretek (egy akkumulátorral)

10.0 x 7.3 x 3.1 in (25.4 x 18.5 x 7.9 cm)

Méretek (kettős akkumulátorral)

11.0 x 7.3 x 3.1 in (27.9 x 18.5 x 7.9 cm)

Tömeg (egy akkumulátorcsomaggal)

2,3 kg (5 lbs)

Tömeg (két akkumulátorcsomag esetén)

2,7 kg (6 lbs)

Névleges zajszint***Áramlási beállítások
Kimeneti O2 beállításai([20 lélegzet/perc STPD mellett]Minden térfogat +/- 15%)
Oxigénkoncentráció*Oxigénmonitor
Maximális kimeneti nyomás
Adagolás érzékenysége

39,93 dB(A) ± 0,42 dB(A) @ 2
Impulzusbeállítások 1, 2, 3, 4, 5
1. beállítás – 210 ml/perc
2. beállítás – 420 ml/perc
3. beállítás – 630 ml/perc
4. beállítás – 840 ml/perc
5. beállítás – 1050 ml/perc
90% (+5.5% / -3%)
Igen
< 30 PSI> -0,1 H2O cm (beállítható)

Váltóáram

100–240 V váltakozó áram, 50–60 Hz, legfeljebb 120 W

Egyenáram

11–18 VDC (max. 10 A)

Akkumulátor típusa

Lítiumion

Akkumulátor kapacitása

Egy cellás: 6700 mAh; Dupla cellás: 13 400 mAh

Akkumulátorfeltöltési idő

Egy akkumulátor esetén: 3,5 óra
Két akkumulátor esetén: 6,0 óra

Akkumulátorcsomag üzemidejeÜzemi hőmérséklet**Egy akkumulátor üzemideje: 2. beállítás – legfeljebb 4 óraKettős akkumulátor üzemideje: 2. beállítás – legfeljebb 8 óra
5 °C – 40 °C (41 °F – 104 °F)

Szállítási és tárolási hőmérséklet (egység)-25 °C – +70 °C (-13 °F – +158 °F) legfeljebb 3048 m (10 000 ft)
magasságig

Szállítási és tárolási hőmérséklet (akkumulátor)-25 °C – +70 °C (-13 °F – +158 °F) legfeljebb 3048 m (10 000 ft)
magasságigÜzemeltetési magasság**Üzemeltetési páratartalom

Tárolási páratartalom
Korlátozott garancia-381–3048 m (-1250–10 000 láb) (700–1060 hPa-ra tesztelve)
15–95% relatív páratartalom (nem lecsapódó)
0−90%, nem lecsapódó
Lásd a jótállási nyilatkozatot* 101 kPa (14,7 psi) légköri nyomás alapján, 21 °C-on (70 °F)** Az egység minden környezetben működni fog, feltéve, hogy a környezet megfelel a megadott hőmérséklet- és
páratartalom-tartománynak. A megadott üzemeltetési körülményeken kívül történő üzemeltetés esetén nagyobb áramlási
sebesség mellett a koncentrátor csak korlátozottan tud megfelelni az oxigénkoncentrációs jellemzőknek. A bemelegedési
képernyő látható, amíg a koncentrátor el nem éri a célkoncentrációt, vagy legfeljebb 15 percig. Használja tovább az
egységet ezalatt az idő alatt.
*** A 14-1 10/2018 MDS-Hi vizsgálati módszer szerint mért zajszint.
A várt üzemi élettartam legalább öt év.

522- HU PN MN232-C4 L | Felhasználói kézikönyv

FreeStyle Comfort
Kezelési utasítások
1. Keressen egy megfelelő helyet a FreeStyle Comfort
használatához.
a. A koncentrátornak megfelelő légáramlásra
van szüksége, így a légbeszívó szűrőket és alégkimeneti nyílást nem szabad eltorlaszolni.
b. A koncentrátort úgy kell elhelyezni, hogy afény- és hangjelzéseket, valamint a riasztásokat
könnyen észlelhesse.
c. Mielőtt bekapcsolná az egységet, ellenőrizze,
hogy a légbeszívó szűrők a helyükön
vannak-e. Ha az egységből hiányzik egy szűrő,akkor vegye fel a kapcsolatot a berendezés
forgalmazójával.

3. Csatlakoztassa a váltó- vagy az egyenáramú
tápegységet az elülső oldal jobb alsó részén lévőelektromos aljzathoz.

2. Helyezzen be egy akkumulátorcsomagot akoncentrátorba.
a. Igazítsa az akkumulátort a FreeStyle Comfort
burkolata alsó részének középvonalához.

4. Csatlakoztassa az orrkanült az oxigénkimenethez.
Használat előtt ellenőrizze a csatlakoztatást.
MEGJEGYZÉS: Az ajánlottakon kívüli kanül(ök)
használata hatással lehet a teljesítményre.

Megjegyzés: A berendezés forgalmazójának: A következő
oxigénadagoló tartozékok javasoltak a FreeStyle Comfort
eszközzel való használatra:b. Csúsztassa befelé az akkumulátort, amíg
kattanást nem hall, amely azt jelzi, hogy az
akkumulátor biztosan rögzül a helyén.

• Orrkanül 2,1 m (7 láb) hosszú csővel (maximális LPM:
6 LPM): CAIRE cikkszám: 5408-SEQ• További ajánlott tartozékokra vonatkozó információ
elérhető online: www.caireinc.com.
5. A főkapcsolóval kapcsolja be a FreeStyle Comfort
készüléket, és válassza ki az előírt áramlási
sebességet az áramlási sebességet növelő (+) vagy
csökkentő (-) gombot használva.

PN MN232-C4 L | Felhasználói kézikönyv HU - 523

FreeStyle Comfort
Légzésszám
15
20
25
30
35
40
Egy rövid hangjelzés jelzi, hogy a készüléket
bekapcsolta.

1

nem autoSAT impulzustérfogat (ml)*beállítása
2
3
4
5

14.0
10.5
8.4
7.0
6.0
5.3

28.0
21.0
16.8
14.0
12.0
10.5

42.0
31.5
25.2
21.0
18.0
15.8

56.0
42.0
33.6
28.0
24.0
21.0

70.0
52.5
42.0
35.0
30.0
26.3

8. A főkapcsoló megnyomásával tudja kikapcsolnia FreeStyle Comfort készüléket.

Megjegyzés: Amikor először kapcsolja be a FreeStyle
Comfort készüléket, az áramlási sebesség alapértelmezett beállítása 2. Ha a beállítást megváltoztatja, akkor az
egység bekapcsolásakor a beállított áramlási sebesség
lesz érvényes.
6. A kanül gyártójának utasításai szerint helyezze
az arcára az orrkanült, és lélegezzen normálisan.
Minden belégzéskor érezni fogja, ahogy egy adag
oxigén jut az orrába az orrkanülön át.
7. Ha meg akarja változtatni az áramlási sebességet,akkor nyomja meg az áramlási sebességet
növelő (+) vagy csökkentő (-) gombot, amíg elnem éri a kívánt beállítást. Természetes, hogy azáramlási sebesség megváltoztatásával a hang isváltozik.

Megjegyzés: A kijelző készenléti állapotba vált, és
20 másodperc múlva elsötétül. Készenléti állapotbanbármelyik gombot megnyomva visszahozza a képernyőt
a készenléti állapotból.
9. A speciális funkciókat lásd a Menü navigáció
utasításoknál.

A TERMÉK SPECIÁLIS FUNKCIÓI
autoSAT®
A CAIRE szabadalmaztatott autoSAT funkciója
lehetővé teszi a FreeStyle Comfort számára, hogyállandó mennyiségű oxigént (bólust) biztosítson
lélegzetenként, a maximális légzési sebességig.
Amikor a felhasználó légzésszáma növekszik olyan
tevékenységek során, mint a séta, az autoSAT funkció
úgy állítja be az eszköz működését, hogy fenntartsa abeállított bólusmennyiséget.
UltraSense®
Ez a szenzitív, impulzusadagolást elindító érzékelő
figyeli a légzés során bekövetkező nyomásváltozást,és a belégzéskor adagolja az oxigénimpulzust.

Légzésszám
15
20
25
30
35
40

autoSat impulzustérfogat (ml) *beállítása
1
2
3
4
5
10.5
10.5
10.5
10.5
10.5
10.5

21.0
21.0
21.0
21.0
21.0
21.0

31.5
31.5
31.5
31.5
30.0
26.3

42.0
42.0
42.0
35.0
30.0
26.3

*Tűrés: +/- 15%524- HU PN MN232-C4 L | Felhasználói kézikönyv

52.5
52.5
42.0
35.0
30.0
26.3

autoDOSE®
Ha „nincs légzés riasztás” következne be, ez a funkció
adagolja az oxigént 20/perces légzésszámmal az
egység aktuális áramlási sebesség beállítása mellett,amíg a riasztási állapot meg nem szűnik.

FreeStyle ComfortÁramellátás
A FreeStyle Comfort három módon kaphat áramot:• Tölthető akkumulátorcsomag• Váltóáram
• Egyenáram
Az egyes módokhoz kapcsolódó működtetésről
a következő részekben kap tájékoztatást.
FIGYELMEZTETÉS: CSAK AZ EBBEN A
FELHASZNÁLÓI KÉZIKÖNYVBEN LEÍRT
ÁRAMELLÁTÁSI MÓDOKAT HASZNÁLJA.
A NEM CAIRE GYÁRTMÁNYÚ TÁPEGYSÉGEK ÉS AKKUMULÁTORCSOMAGOKHASZNÁLATA KÁROSÍTHATJA AZ
EGYSÉGET, ÉS HELYTELEN MŰKÖDÉST
EREDMÉNYEZHET.

AZ AKKUMULÁTOR ELTÁVOLÍTÁSA

Nyomja a FreeStyle Comfort készülékhez képest lefelé
az akkumulátor kioldógombját. Miközben benyomva
TARTJA ezt a gombot, csúsztassa ki az akkumulátort
úgy, hogy kihúzza azt a koncentrátorból.
FIGYELMEZTETÉS: AZOXIGÉNKONCENTRÁTORT KI KELL
KAPCSOLNI ÉS KI KELL HÚZNI AFALI CSATLAKOZÓBÓL, MIELŐTT AZ
AKKUMULÁTORCSOMAGOT KIVESZI.

AKKUMULÁTORCSOMAG

A FreeStyle Comfort működtethető a készülékhez
mellékelt tölthető akkumulátorcsomagról. A koncentrátorműködése nincs korlátozva, ha az akkumulátorcsomagrólműködik. Minden áramlási sebesség a rendelkezésre áll
akkumulátoros áramellátás esetén.

AZ AKKUMULÁTORCSOMAG
TÖLTÖTTSÉGÉNEK ELLENŐRZÉSEAZ AKKUMULÁTOR BEHELYEZÉSE

Igazítsa az akkumulátort a FreeStyle Comfort
burkolata alsó részének középvonalához. Kattanásig
nyomja befelé az akkumulátort, hogy az biztosanrögzüljön a helyén.

A FreeStyle Comfort készülékbe behelyezett
akkumulátor töltöttségét az akkumulátor ikon ésa becsült hátralévő működési idő mutatja a kijelzőn.
80–100%60–80%

40–60%

20–40%

20%

PN MN232-C4 L | Felhasználói kézikönyv HU - 525

FreeStyle Comfort
Megjegyzés: Ha áramforrásra van csatlakoztatva azeszköz, dugasz jelzés jelenik meg a töltöttségi állapot
százalékos kijelzése helyett.
Ha az akkumulátor töltöttségét úgy kívánja ellenőrizni,
hogy az akkumulátor nincs benne a FreeStyle
Comfort készülékben, akkor helyezze azt vízszintes
felületre, és nyomja meg a TEST (Teszt) gombot.
Az akkumulátor töltöttségét jelző fények (25–100%)
kigyulladnak a TEST (Teszt) gomb jobb oldalán,mutatva az akkumulátorcsomag töltöttségi szintjét.

AZ AKKUMULÁTOR MŰKÖDÉSI IDEJE

Ha lecsatlakoztatta a külső váltó- vagy egyenáramú
áramforrásról, akkor a koncentrátor automatikusan
az akkumulátor használatára vált, ha be van helyezve
egy feltöltött akkumulátor. A lenti táblázatban egy új
akkumulátorcsomag jellemző működési ideje látható.
Az akkumulátorcsomag működési ideje állandó,
függetlenül a légzésszámtól.
Impulzusmód
beállítás

Egyes (8 cellás)
akkumulátorműködési idejeKettős (16 cellás)
akkumulátorműködési ideje

1

legfeljebb 8 óra

legfeljebb 16 óra

2

legfeljebb 4 óra

legfeljebb 8 óra

3

legfeljebb 3 óra

legfeljebb 6 óra

4legfeljebb 2,25 óra

legfeljebb 4,5 óra

5legfeljebb 2 óra

legfeljebb 4 óra

AZ AKKUMULÁTOR JELLEMZŐ
FELTÖLTÉSI IDEJEELSŐ TÖLTÉS

A FreeStyle Comfort készülékhez mellékelt új
akkumulátor csak részlegesen van feltöltve, amikora gyárból kiszállítják. Az első használat előtt váltóvagy egyenáramú áramforrásból töltse fel teljesenaz akkumulátorcsomagot.

ÁLTALÁNOS TUDNIVALÓK A TÖLTÉSRŐL
A FreeStyle Comfort készülék mindig tölti az
akkumulátorcsomagot, ha az be van helyezve a
koncentrátorba, és a koncentrátor külső váltó- vagy
egyenáramú áramforrásra van csatlakoztatva(kivéve a repülőn).
Ha az akkumulátor teljesen fel van töltve, akkor
a kijelzőn a 100% jelzés látható.
Alternatívaként az akkumulátorcsomag az opcionális
asztali töltővel is tölthető.

526- HU PN MN232-C4 L | Felhasználói kézikönyv

Az akkumulátorcsomag jellemző teljes feltöltési ideje
teljesen lemerült akkumulátor esetén 3,5 óra egyes
akkumulátornál, és 6,0 óra kettős akkumulátornál.
Megjegyzés: Az akkumulátorcsomag teljes lemerülése
esetén bizonyos esetekben az akkumulátorcsomagelőször nem kezd töltődni.

AZ AKKUMULÁTOR ÜZEMI
ÉLETTARTAMA

A FreeStyle Comfort akkumulátorokat 300 feltöltési/
lemerülési ciklusra tervezték.

FreeStyle Comfort
Váltóáramú és egyenáramú
tápegység
A FreeStyle Comfort készüléket mindig váltó- és
egyenáramú tápegységgel együtt szállítják.
A váltóáramú tápegység az alábbiakat tartalmazza:	a. Váltóáramú adapter, amely 100–240 volt
feszültségen és 50–60 Hz frekvencián működik.
	b. Váltóáramú tápkábel, amely összeköti az
adaptert és a megfelelő csatlakozóaljzatot.

Az egyenáramú tápegység egy kábelt tartalmaz, amely
közvetlenül köti össze az egységet és az egyenáramú
aljzatot.

2. Csatlakoztassa a váltóáramú tápkábelt a váltóáramú
tápegységhez.

VÁLTÓÁRAM

Váltóáramot használnak az otthonokban, ésbárhol, ahol a szabványos váltóáramú áramellátás
rendelkezésre áll. A FreeStyle Comfort
teljes mértékben működőképes váltóáramú
áramellátás esetén. A FreeStyle Comfort tölti
is az akkumulátorcsomagot, ha váltóáramra
van csatlakoztatva (kivéve a repülőn). Az
akkumulátorcsomag töltése független attól, hogy a
FreeStyle Comfort be van-e kapcsolva vagy sem.
FIGYELMEZTETÉS: NE CSATLAKOZTASSA
KÜLSŐ ÁRAMFORRÁSHOZ, HA
AZ AKKUMULÁTORCSOMAG
NINCS BEHELYEZVE AZOXIGÉNKONCENTRÁTORBA.

3. A váltóáramú tápkábel másik végét csatlakoztassa
a fali vagy egyéb, megfelelő elektromos aljzathoz.

A FreeStyle Comfort csatlakoztatása váltóáramú
áramforráshoz:
1. Csatlakoztassa a váltóáramú tápegységet a
FreeStyle Comfort elülső oldalának jobb alsó részén
található áramcsatlakozóhoz. Jól nyomja bele, hogy
biztosan csatlakoztassa.

PN MN232-C4 L | Felhasználói kézikönyv HU - 527

FreeStyle Comfort
4. A főkapcsolóval kapcsolja be a FreeStyle Comfort
készüléket, és válassza ki az előírt áramlási
sebességet az áramlási sebességet növelő (+) vagy
csökkentő (-) gombot használva.

4. A főkapcsolóval kapcsolja be a FreeStyle Comfort
készüléket, és válassza ki az előírt áramlási
sebességet az áramlási sebességet növelő (+) vagy
csökkentő (-) gombot használva.
Megjegyzés: Ha tölteni kell az akkumulátorcsomagot,
akkor a kijelzőn az akkumulátor ikonon egyetlen narancssárga sáv látható.

EGYENÁRAM

A készülék egyenárammal az autók, a hajók és egyéb
gépjárművek elektromos csatlakozóiról használható.
A FreeStyle Comfort teljes mértékben működőképes
egyenáramú áramellátás esetén. A FreeStyle Comfort
szükség esetén tölti is az akkumulátorcsomagot, haegyenáramot kap. Az akkumulátorcsomag töltése
független attól, hogy az egység be van-e kapcsolva
vagy sem.
FIGYELMEZTETÉS: NE CSATLAKOZTASSA
KÜLSŐ ÁRAMFORRÁSHOZ, HA
AZ AKKUMULÁTORCSOMAG
NINCS BEHELYEZVE AZOXIGÉNKONCENTRÁTORBA.
1. Indítsa be a gépjárművet.
2. Csatlakoztassa az egyenáramú tápegységet a
FreeStyle Comfort elülső oldalának jobb alsó részén
található áramcsatlakozóhoz. Jól nyomja bele, hogy
biztosan csatlakoztassa.

3. Az egyenáramú tápkábel másik végét csatlakoztassa
az autó vagy más jármű egyenáramú aljzatához.

528- HU PN MN232-C4 L | Felhasználói kézikönyvVIGYÁZAT: Az esetleges tűz vagy
túlmelegedés megelőzése érdekébengyőződjön meg arról, hogy a 12 V-os
egyenáramú tápegység csatlakozóaljzata
nem tartalmaz törmeléket, és az egyenáramú
tápegység adaptercsatlakozója megfelelően
illeszkedik és biztonságosan a helyén marad.
VIGYÁZAT: Az egyenáramú tápegység fémfelületeia nagy áramerősség miatt használat közben NAGYONFELFORRÓSODHATNAK. Érintés előtt hagyjon
elegendő időt a felületek lehűlésére.
VIGYÁZAT: Ne használja az egyenáramú tápegységet
hosszabbítókkal, osztókkal vagy adapterekkel. A nem
jóváhagyott tartozékok használata túlmelegedést vagykárosodást okozhat.
VIGYÁZAT: Ha az eszközt egy 12 V-os egyenáramú
aljzathoz csatlakoztatja, akkor a jármű akkumulátora
lemerülhet, ha a motor nem jár, még akkor is, ha azeszköz ki van kapcsolva.
VIGYÁZAT: Ne indítsa el a járművet, amíg azegyenáramú tápegység csatlakoztatva van.

FreeStyle Comfort
Tisztítás, ápolás és rendszeres
karbantartás
RENDSZERES KARBANTARTÁS

Megelőző karbantartás
Az eszköz karbantartást nem igényel. Az eszköz
belsejében lévő szűrők az eszköz élettartama alattműködnek. Ha az eszköz nem működik megfelelően,az eszközt vissza kell juttatni a forgalmazóhoz
javításra. A FreeStyle Comfort házán belüli belső
alkatrészek szervizelését csak a készülék forgalmazója
végezheti.
Az eszköz használója az alábbi karbantartásiműveletekért felelős.
• Hetente – a légbeszívó szűrők tisztítása.
FIGYELMEZTETÉS: NE PRÓBÁLJA MEG
FELNYITNI VAGY ELTÁVOLÍTANI AZOXIGÉNKONCENTRÁTOR MŰANYAGBURKOLATAIT. AZ EGYSÉGET CSAKSZAKKÉPZETT SZERVIZTECHNIKUS
NYITHATJA FEL.

LÉGBESZÍVÓ SZŰRŐK

A levegő a FreeStyle Comfort légbeszívó szűrőin
keresztül jut be az oxigénkoncentrátor elülső részén,alul. Ezek a szűrők akadályozzák meg, hogy a por
vagy más nagyobb részecskék az egységbe jussanak.
Hetente legalább egyszer meg kell tisztítani alégbeszívó szűrőket. A forgalmazó azt tanácsolhatja,
hogy az üzemeltetési körülményektől függően akár
gyakrabban is végezze el a tisztítást. A légbeszívószűrők helyes megtisztítása érdekében kövesse az
alábbi lépéseket.
1. A FreeStyle Comfort elülső oldalából távolítsa el a
két szűrőt úgy, hogy megnyomja a megfelelő füleket
és kihúzza a szűrőket a készülékből.

FIGYELMEZTETÉS: NE MŰKÖDTESSE AZOXIGÉNKONCENTRÁTORT BEHELYEZETTLÉGBESZÍVÓ SZŰRŐ NÉLKÜL!

AZ AKKUMULÁTORCSOMAGGONDOZÁSA
Ne hagyja, hogy nedvesség érje az
akkumulátorcsomagot!

Ne próbálja meg szétszerelni vagy kinyitni az
akkumulátorcsomagot!
Ne tegye ki az akkumulátort nyílt lángnak!
Ne ejtse le az akkumulátorcsomagot!
Tartsa gyermekektől távol az akkumulátorcsomagot!
Ha az akkumulátorcsomag sérültnek tűnik, akkor
használat előtt vegye fel a kapcsolatot a forgalmazóval.
Ne tegye az akkumulátort mikrohullámú sütőbe vagy
egyéb konyhai készülékbe, illetve ezek közelébe!

AZ AKKUMULÁTORCSOMAG TÁROLÁSA
Töltés előtt nem kell teljesen lemeríteni az
oxigénkoncentrátor akkumulátorát. Minden használat
után ajánlott feltölteni az oxigénkoncentrátor
akkumulátorát.

Ne tegye olyan helyre az akkumulátort, ahol nagyhőhatásnak van kitéve, pl. közvetlen napfényre vagy
egy autóba. A hő rontja az akkumulátor teljesítményét
és az élettartamát is. Hő hatására az akkumulátor is hőt
termelhet, füstölhet, kigyulladhat vagy felrobbanhat.
Az akkumulátorcsomagot állandó környezeti
körülmények között, alacsony páratartalomban(kevesebb, mint 70% relatív páratartalom), korrozív
gázoktól mentes helyen kell tárolni. A rozsdásodásmegelőzése érdekében kerülje a páralecsapódásra
hajlamosító körülményeket, például a környezetihőmérséklet gyors ingadozását.
Egy hónapos vagy annál hosszabb tároláshoz alemerült vagy 50% és 75% közötti (két vagy három
LED) részlegesen töltött állapot ajánlott. Ha feltöltötte,vegye ki az akkumulátort az oxigénkoncentrátorból.

2. Mindkét szűrőt meleg, szappanos vízben mossa át.
3. Alaposan öblítse le a szűrőket, majd puha,szálmentes törlőkendővel törölje le a felesleges
vizet. Ellenőrizze, hogy a szűrők teljesen
megszáradtak-e, mielőtt visszahelyezi azokat.
4. Helyezze vissza a tiszta, száraz szűrőket.

PN MN232-C4 L | Felhasználói kézikönyv HU - 529

FreeStyle Comfort
TISZTÍTÁS

Kapcsolja KI az egységet, és tisztítás előtt válasszale a váltóáramú vagy egyenáramú feszültségről.
SEMMIT NE permetezzen közvetlenül a külső
burkolatra. Használjon nedves (nem átitatott) kendőt
vagy szivacsot. A törlőkendőre vagy szivacsra
permetezzen enyhe hatású háztartási tisztítószeroldatot
a burkolat és a tápegység megtisztításához. Folytassa atisztítást a tisztítószer gyártójának utasításai alapján.
FIGYELMEZTETÉS: AZ EGYSÉG
TISZTÍTÁSA ELŐTT KAPCSOLJAKI AZT, ÉS A TÁPKÁBELT HÚZZAKI AZ ELEKTROMOS ALJZATBÓL,HOGY ELKERÜLJE AZ ÁRAMÜTÉST
VAGY AZ ÉGÉSI SÉRÜLÉST. CSAK ABERENDEZÉS FORGALMAZÓJA VAGYEGY SZAKKÉPZETT SZERVIZTECHNIKUS
TÁVOLÍTHATJA EL A BURKOLATOKAT,
VAGY SZERVIZELHETI AZ EGYSÉGET.
FIGYELMEZTETÉS: ÓVATOSAN KELL ELJÁRNI,HOGY AZ OXIGÉNKONCENTRÁTORT ÉS AZAKKUMULÁTORCSOMAGOT NE ÉRJE NEDVESSÉG,
VALAMINT NE KERÜLJÖN FOLYADÉK AZ EGYSÉGBE.
EZ HIBÁS MŰKÖDÉST VAGY ÜZEMSZÜNETET
EREDMÉNYEZHET, ÉS MEGNŐ AZ ÁRAMÜTÉS VAGYAZ ÉGÉSI SÉRÜLÉS KOCKÁZATA.
FIGYELMEZTETÉS: NE HASZNÁLJON FOLYADÉKOTKÖZVETLENÜL AZ EGYSÉGEN! A MELLŐZENDŐVEGYI ANYAGOK LISTÁJA TÖBBEK KÖZÖTT, DE NEMKIZÁRÓLAGOSAN AZ ALÁBBIAKAT TARTALMAZZA:ALKOHOL ÉS ALKOHOL ALAPÚ TERMÉKEK,KONCENTRÁLT KLÓR ALAPÚ TERMÉKEK(ETILÉN-KLORID), ÉS OLAJ ALAPÚ TERMÉKEK.
EZEK NEM HASZNÁLHATÓAK A KONCENTRÁTOR
MŰANYAGBURKOLATÁNAK TISZTÍTÁSÁRA, MIVELKÁROSÍTHATJÁK AZ EGYSÉGET.
FIGYELMEZTETÉS: A KÉSZÜLÉK HÁZÁNAK, KEZELŐPULTJÁNAK ÉS TÁPKÁBELÉNEK TISZTÍTÁSÁRA
CSAK ENYHE HÁZTARTÁSI TISZTÍTÓSZERREL
BENEDVESÍTETT (NEM ÁTITATOTT) KENDŐT VAGYSZIVACSOT HASZNÁLJON, MAJD TÖRÖLJÖN
SZÁRAZRA MINDEN FELÜLETET. ÜGYELJEN ARRA,NEHOGY FOLYADÉK KERÜLJÖN A KONCENTRÁTORBA. FORDÍTSON KÜLÖNÖS FIGYELMET A KANÜLHÖZVEZETŐ OXIGÉNKIMENETI CSATLAKOZÓRA, HOGYAZ MENTES MARADJON MINDEN PORTÓL, VÍZTŐLÉS SZEMCSÉTŐL.

530- HU PN MN232-C4 L | Felhasználói kézikönyv

FIGYELMEZTETÉS: EGY NEDVES (NEMÁTITATOTT) TÖRLŐKENDŐT VAGY SZIVACSOTHASZNÁLJON AZ AKKUMULÁTORCSOMAG
MEGTISZTÍTÁSÁHOZ. ELŐSZÖR PERMETEZZENHÁZTARTÁSI TISZTÍTÓSZERT A TÖRLŐKENDŐRE
VAGY A SZIVACSRA, ÉS EZZEL TISZTÍTSA MEG
AZ AKKUMULÁTORCSOMAG BURKOLATÁT.
FIGYELMEZTETÉS: NE TISZTÍTSA AUTOMATAMOSÓBERENDEZÉSBEN!

A KANÜL CSERÉJE

Mindig kövesse a kanül gyártójának a helyes
használatra vonatkozó utasításait. 3–6 havonta cseréljea kanült és az oxigénvezetéket (napi használat esetén
3 havonta) a gyártó ajánlásai szerint, a orrkanült
gyártójának vagy a készülék forgalmazójának az
ajánlásai szerint. Az orvos vagy az oxigénkoncentrátor
forgalmazója ellátja Önt a tisztításra és a cserérevonatkozó utasításokkal.
A cseréhez szükséges plusz kellékek beszerezhetők
az oxigénkoncentrátor forgalmazójától.

FreeStyle Comfort
Riasztási körülmények
Fontossági
sorrend

Riasztásleírása1

Rendszermeghibásodás,
akkumulátor
magas hőmérséklet

Kijelző

Hangjelzés

További részletek/Javasolt művelet

1 sípolás
20 másodpercenként

Az egység riasztást ad ki, ha az akkumulátorhőmérséklete megközelíti a biztonsági lekapcsolás határértékét.
Ha eléri a lekapcsolási határértéket, az egység
kikapcsol. Ha külső áramforrásra van csatlakoztatva, a riasztási képernyő jelenik meg, és ariasztási hangjelzés 2 percig hallható.
Szobahőmérsékletű akkumulátorral tesztelje.
Ha a probléma továbbra is fennáll, forduljon
a forgalmazóhoz.

2

Rendszermeghibásodás, magashőmérséklet

1 sípolás
20 másodpercenként

Az egység riaszt, ha a belső hőmérséklet
megközelíti a lekapcsolás határértékét.
Ha eléri a lekapcsolási határértéket, az
egység kikapcsol, megjelenik a riasztási
képernyő, és a riasztási hangjelzés 2 percig
hallható.
Ellenőrizze, hogy semmi nem zárja el az
egység légkimeneti nyílásait. Ha a probléma
továbbra is fennáll, forduljon a forgalmazóhoz.

3

Ha eléri a lekapcsolási határértéket, az
egység kikapcsol, megjelenik a riasztási
képernyő, és a riasztási hangjelzés 2 percig
hallható.

Rendszermeghibásodás, magas
nyomás(magasrendszernyomás vagy
kompresszorhiba)

1 sípolás
20 másodpercenként

4

Rendszermeghibásodás, áramlási
sebesség(OCD szelep
meghibásodása, légvezeték
eltömődése)

1 sípolás
30 másodpercenként

Ha a probléma továbbra is fennáll, forduljon
a forgalmazóhoz.

5

Akkumulátor
lemerül,lekapcsolás

1 sípolás
30 másodpercenként

Riasztás, amíg az akkumulátor le nem merülés az egység ki nem kapcsol.

Ha a probléma továbbra is fennáll, forduljon
a forgalmazóhoz.

Töltse az akkumulátort.

PN MN232-C4 L | Felhasználói kézikönyv HU - 531

FreeStyle Comfort
Fontossági
sorrend

Riasztásleírása6

Nemészlelhető
légzés

Kijelző

Hangjelzés

További részletek

1 sípolás
10 percenként

15 másodperce nem észlel légzést.
Az első légzés, vagy a bemelegedési
képernyő lekapcsolása aktiválja, amelyikelőbb következik be.
Ellenőrizze, hogy a kanül nem tört-e meg,vagy nincs-e rajta repedés. Állítsa nagyobb
fokozatra az érzékenységet. Ha a probléma
továbbra is fennáll, forduljon a forgalmazóhoz.

7

Bemelegedésnincs

Az egység megkezdi az automatikus
adagolást, amíg légzést nem észlel, vagy
a bemelegedés véget nem ér, amelyik előbb
következik be.

8

Alacsony
koncentráció

1 sípolás
10 percenként
maximum
30 percig

Ha a probléma továbbra is fennáll, forduljon
a forgalmazóhoz.

9

Akkumulátor
lemerül figyelmeztetés

1 sípolás,
nincs
ismétlés

Töltse az akkumulátort.

10

Légzésszámtúllépés

1 sípolás,
nincs
ismétlésA légzésszám meghaladja a 40/perc értéket.

11

Rendszermeghibásodás,
ventilátorhiba

nincs

Ha a probléma továbbra is fennáll, forduljon
a forgalmazóhoz.

12

Rendszermeghibásodás, alacsony
nyomásnincs

Ha a probléma továbbra is fennáll, forduljon
a forgalmazóhoz.

13

Rendszermeghibásodás, alacsonyhőmérséklet

nincs

Várja meg, amíg az egység szobahőmérsékletre melegszik. Ha a probléma továbbra isfennáll, forduljon a forgalmazóhoz.

532- HU PN MN232-C4 L | Felhasználói kézikönyv

Ha az adagolást téves növelése történt, akkor
csökkentse az érzékenységet. Ha a probléma
továbbra is fennáll, forduljon a forgalmazóhoz.

FreeStyle Comfort
Fontossági
sorrend

Riasztásleírása

Tájékoztató

Indítás

Kijelző

Hangjelzés

További részletek

1 tájékoztató sípolás

A képernyő megjelenik bekapcsoláskor,és megszólal egy sípolás, amely jelzi,
hogy a hangszóró működik.
Az indulási képernyőn megjelenik a firmware
aktuális verziója a jobb alsó sarokban.

Tájékoztató

Normál működési képernyő
autoSATüzemmódban (nincs
egyenáramú
tápellátás
csatlakoztatva)

nincs

Tájékoztató

Normál működési képernyő
autoSATüzemmód
letiltva (nincs
egyenáramú
tápellátás csatlakoztatva)

nincs

Tájékoztató

Beállítás megváltoztatva

1 tájékoztató sípolás

Tájékoztató

Kikapcsolt állapotban külső
áramforrásra
csatlakozott

nincs

TájékoztatóMűködés
közbencsatlakoztatjakülsőáramforrásra

nincs

Tájékoztató

Kikapcsoltállapotban
leválasztja
a külső
áramforrásról

1 tájékoztató sípolás

2 másodpercre megjelenik a CAIRE képernyő,amit 5 másodpercig az akkumulátor és acsatlakozó állapota követ, mielőtt kikapcsol
a képernyő.

Megjelenik a CAIRE képernyő 2 másodpercre,
majd a kijelző kikapcsol.

PN MN232-C4 L | Felhasználói kézikönyv HU - 533

FreeStyle Comfort
Fontossági
sorrend

Riasztásleírása

Kijelző

TájékoztatóMűködés
közbenleválasztja akülső áramforrásról (azakkumulátor
behelyezve)

1 tájékoztató sípolás

Tájékoztató

Készenléti
módban –ha a fel vagy
le gombot
megnyomja,
a képernyőn
megjelenik az
akkumulátor
töltöttsége
és a töltés
jelzésenincs

Tájékoztató

Repülőgép
üzemmód
akkumulátorról működve

1 hosszúsípolás

Tájékoztató

Repülőgép
üzemmód
hálózatra
csatlakoztatva

1 hosszúsípolás

534- HU PN MN232-C4 L | Felhasználói kézikönyv

Hangjelzés

További részletek

A képernyő 5 másodpercre megjelenik,
majd lekapcsol.

FreeStyle Comfort
Menünavigációs utasítások
Készenléti vagy Futtatási üzemmódban tud a menübe
belépni. Ha Futtatási üzemmódban riasztási állapot
következik be, akkor automatikusan kilép a menüből.
Ha 20 másodpercig semmilyen gombot sem nyom
meg, akkor automatikusan kilép a menüből. Készenlétiüzemmódban lehet, hogy kétszer is meg kell nyomnia
a Menü gombot, hogy felébressze az egységet, és betudjon lépni a menübe.
Négy menüképernyő van:
1. Állapot képernyő: megjelenik a firmware-verzió
száma, a futtatási idő és az akkumulátor lemerülési
ciklusainak a száma

Repülőgép üzemmód

2. Repülőgép üzemmód (BE/KI)
3. Érzékenység: állítható (1–5)
4. autoSAT (BE/KI)

ÉrzékenységÁllapot képernyő példaBEÁLLÍTÁSOK MENÜ MŰKÖDÉSE
A Beállítások menübe a Menu/Select (Menü/Kiválasztás)
gomb
megnyomásával tud lépni Futtatási ésKészenléti módban.
A menük végiggörgetéséhez nyomja meg a

autoSAT

gombot.

Állapot lap

PN MN232-C4 L | Felhasználói kézikönyv HU - 535

FreeStyle ComfortÁLLAPOT LAP

A Repülőgép üzemmód törléséhez nyomja meg

A FreeStyle Comfort Állapot lapja mutatja a firmwareverzió számát, a futtatási időt és az akkumulátorciklusok
számát a csatlakoztatott akkumulátoron.

agombot, ha a képernyőn a CANCEL (TÖRLÉS)
ikon látható.

Repülőgép üzemmód BE megerősítveÁllapot lap

A futtatási idő egész órák kifejezve jelenik meg.
Az Akkumulátor lemerülései a csatlakoztatottakkumulátor lemerült és újratöltött ciklusainak száma.

REPÜLŐGÉP ÜZEMMÓD
A FreeStyle Comfort kikapcsolt Repülőgép
üzemmóddal kerül kiszállításra.

A Repülőgép mód kikapcsolásához ismételje meg
a fentieket, míg az OFF (KI) szóra vált.
A
gombot megnyomva, miközben a képernyőn
a „Next” (Következő) szimbólum látható, a menü
oldalt a következő képernyő opcióra váltja.
A Repülőgép üzemmód Futtatás üzemmódban
való használatra szolgál. A Repülőgép üzemmód
automatikusan letiltásra kerül, ha az egység
akkumulátorról működik, és Készenléti módban hagyta.
Megjegyzés: Lásd a Repülőgép üzemmód aktiválása részt,ha többet kíván megtudni a Repülőgép üzemmódról.

ÉRZÉKENYSÉG MÓD

Repülőgép üzemmód főképernyője

Az üzemmód bekapcsolásához nyomja meg agombot, ha a képernyőn a CHANGE (MÓDOSÍTÁS)ikon látható. A kijelzőn látható szó kék „ON” (BE)szóra vált.

A FreeStyle Comfort eszközt a 2-es érzékenység gyári
beállítással szállítják, és normál körülmények között
ezt nem szükséges módosítani. Az érzékenységet 1 és
5 között lehet beállítani, ahol az 1-es érték a legkisebb,
az 5-ös pedig a legnagyobb érzékenységet jelöli.

Érzékenység mód képernyője 2-es beállítással

Repülőgép üzemmód BE kiválasztva

A Repülőgép üzemmód megerősítéséhez nyomja meg
agombot, ha a képernyőn a SAVE (MENTÉS)
ikon látható. A kijelzőn látható szó fekete „ON” (BE)szóra vált.

536- HU PN MN232-C4 L | Felhasználói kézikönyv

FreeStyle Comfort
Az érzékenység beállításához nyomja meg agombot, ha a képernyőn a CHANGE (MÓDOSÍTÁS)ikon látható. A kijelzőn látható érték a következő kék
számra vált.

3-as érzékenységi mód kiválasztva

A 3. beállítás megerősítéséhez az érzékenység módban
nyomja meg agombot, ha a képernyőn a SAVE
(MENTÉS) ikon látható. A kijelzőn látható érték fekete„3”-ra vált.

AUTOSAT ÜZEMMÓD
A FreeStyle Comfort bekapcsolt autoSAT üzemmóddal
kerül kiszállításra.

autoSAT üzemmód főképernyője

Az üzemmód kikapcsolásához nyomja meg agombot, ha a képernyőn a CHANGE (MÓDOSÍTÁS)ikon látható. A kijelzőn látható szó kék „OFF” (KI)szóra vált.

Az érzékenység törléséhez nyomja meg agombot,ha a képernyőn a CANCEL (TÖRLÉS) ikon látható.

autoSAT üzemmód Ki kiválasztva

Az autoSAT üzemmód kikapcsolásának megerősítéséhez
3-as érzékenységi mód megerősítve

nyomja meg agombot, ha a képernyőn a SAVE
(MENTÉS) ikon látható. A kijelzőn látható szó fekete„OFF” (KI) szóra vált.
Az autoSAT üzemmód bekapcsolásának törléséhez
nyomja meg agombot, ha a képernyőn
a CANCEL (TÖRLÉS) ikon látható.

5-ös (maximális) érzékenységi mód megerősítveA
gombot megnyomva, miközben a képernyőn
a „Next” (Következő) szimbólum látható, a menü
oldalt a következő képernyő opcióra váltja.
autoSAT mód Ki megerősítve

Az autoSAT mód bekapcsolásához ismételje meg
a fentieket, míg az ON (BE) szóra vált.
A
gombot megnyomva, miközben a képernyőn a„Next” (Következő) szimbólum látható, a menü oldalt
a következő képernyő opcióra váltja.

PN MN232-C4 L | Felhasználói kézikönyv HU - 537

FreeStyle Comfort
Hibaelhárítás
Probléma

Lehetséges ok

Megoldás

A FreeStyle Comfort
készüléken megjelenik egy
riasztás és szaggatott sípolás
hallatszik.

Tekintse meg a riasztási
körülmények táblázatát.

Tekintse meg a riasztási körülmények táblázatát.

Az oxigénkoncentrátor nem
kapcsolható be.

Az akkumulátor lemerült.

A váltóáramú vagy az egyenáramú aljzaton
keresztül lássa el az egységet árammal.

Nincs csatlakoztatva a külső
áramforrásra.
Általános meghibásodás.

Ellenőrizze, hogy a külső csatlakozók megfelelően
csatlakoznak-e.
Végezzen teljes ki- és bekapcsolási ciklust úgy,hogy kiveszi az akkumulátort, és leválasztja aváltó- vagy egyenáramú külső áramforrásról.
Ezután helyezze vissza az akkumulátort, és aki- és bekapcsolási ciklus kész. Szükség esetén
csatlakoztassa vissza a külső áramforráshoz.
Forduljon a berendezés forgalmazójához, ésszükség esetén váltson másik oxigénforrásra.

Minden egyéb probléma.

Végezzen teljes ki- és bekapcsolási ciklust úgy,hogy kiveszi az akkumulátort, és leválasztja aváltó- vagy egyenáramú külső áramforrásról.
Ezután helyezze vissza az akkumulátort, és aki- és bekapcsolási ciklus kész. Szükség esetén
csatlakoztassa vissza a külső áramforráshoz.
Váltson másik oxigénforrásra, ha van, és vegye fel
a kapcsolatot a berendezés forgalmazójával.

Megjegyzés: Egyes riasztások miatt az egység leáll.
Megjegyzés: Teljes áramkimaradás esetén a riasztási rendszer alaphelyzetbe áll (minden korábbi riasztás törlődik),és csak az a riasztás jelenik meg, amelynek a feltételei még az áramellátás helyreállása után is fennállnak.

538- HU PN MN232-C4 L | Felhasználói kézikönyv

FreeStyle Comfort
Utazási tudnivalók
A FreeStyle Comfort megfelel a kereskedelmi repülők
fedélzetén történő használattal kapcsolatos FAA
irányelveknek. Minden utazás előtt (repülőn, vonaton,
hajón, stb.) az alábbiakat kell elvégeznie:• Ellenőrizni, hogy a koncentrátor megfelelően
működik-e.
• Az indulás előtt legalább 48 órával vegye fel akapcsolatot az utasszállítóval, és tájékoztassa arról,
hogy FreeStyle Comfort készülékkel fog utazni.
Ajánlott midig tartalék akkumulátort vinni az útra.
Légi utazásnál a földi tennivalókra (repülés előtti
becsekkolás, biztonsági ellenőrzés, stb.) és azátszállásokra is legyen elég az akkumulátor töltöttsége.
Sok légitársaság előírja, hogy az utazás becsült
maximális időtartamának (repülési idő, repülés előtti
és utáni földi tennivalók, csatlakozások, és váratlan
késések) 150%-ára legyen elegendő számú, teljesen
feltöltött akkumulátorcsomag Önnél.

1. CSAK az akkumulátor áramát használva kapcsolja
be a POC-t úgy, hogy az nincs külső áramforrásra
csatlakoztatva.

2. Engedélyezze a Repülőgép üzemmódot az egyik
alábbi módszert használva:a. A Repülőgép üzemmód engedélyezéséhez
kövesse a Menünavigációs utasítások részt.

Megjegyzés: Repülés során a tartalék akkumulátorokat
biztonságos hordtáskában ajánlott tartani.

FAA-SZABÁLYOZÁS

A tartalék akkumulátoroknál külön-külön kellmegakadályozni a rövidzárlat kialakulását úgy,hogy az eredeti gyári csomagolásba rakja vissza
azokat, vagy más módon leszigeteli az érintkezőket,pl. szigetelőszalaggal leragasztja a szabadon lévőérintkezőket, vagy külön nejlonzacskóba vagyvédőtasakba helyezi mindegyik akkumulátort,
és csak a kézi poggyászban viszi magával.

b. Tartsa lenyomva a Fel és a Le gombot
5 másodpercig, amíg egy hosszú sípolást nem hall,és meg nem jelenik a Repülőgép szimbólum.

REPÜLŐGÉP-ÜZEMMÓD AKTIVÁLÁSA
Akkumulátoros áramellátás
A POC a normál működési eljárások szerint
használható repülőgépen CSAK akkumulátorról úgy,hogy nincs külső áramforrásra csatlakoztatva.

3. Ha a repülőgép szimbólum megjelent,
akkor csatlakoztassa a POC-t az aljzathoz.

Külső áramforrásról történő áramellátás
Ha a repülőgépen van elektromos csatlakozóaljzat,
ami működőképes, és megfelelő helyen is van a
POC áramforrásaként, akkor az használható a POC
akkumulátorok biztonsági tartalékaként. Ebben az
esetben az alábbi eljárás végrehajtása a javasolt, hogy
az akkumulátorról a repülőgép áramforrására váltson:
Megjegyzés: A repülőgépre vonatkozó korlátozás miatt az
akkumulátorok nem töltődnek Repülőgép-üzemmódban.

PN MN232-C4 L | Felhasználói kézikönyv HU - 539

FreeStyle Comfort
Tartozékok

AZ EGYSÉG BEHELYEZÉSEA HORDTÁSKÁBA

A megfelelő teljesítmény és a biztonság érdekében
kizárólag az oxigénkoncentrátor forgalmazóján
keresztül a CAIRE által szállított, itt felsorolt
tartozékokat használja. Az alább felsoroltaktól eltérő
tartozékok használata károsan befolyásolhatja akoncentrátor teljesítményét és/vagy a biztonságot.

1. Az egységet a táska aljára helyezze úgy, hogya szűrők és az akkumulátor nyílásai a megfelelő
helyzetben legyenek.

A FreeStyle Comfort szokásos tartozékai
Hordtáska

MI396-1

Hátitáska

MI459-1

VállszíjMI397-1

Váltóáramú tápegység(minden kábellel együtt)

PW036-1S

Egyenáramú tápkábel

CD041-1

Durva részecskék külső szűrője
(jobb oldali)

FI226-1

Durva részecskék külső szűrője
(bal oldali)

FI227-1

2. A táska alján húzza össze a cipzárt.

Opcionálisan rendelhető tartozékok
Kiegészítő 8 cellás akkumulátorcsomag

BT034-1S

16-cellás akkumulátorcsomag

BT035-1S

Asztali akkumulátorcsomag-töltő

BT036-1

Tartozék hordtáska

MI372-2

Megjegyzés: További opciók lehetnek elérhetők
az országspecifikus tápkábelekhez, ahol fent
jelölve van. Forduljon a CAIRE vállalathoz vagy az
oxigénkoncentrátor forgalmazójához, ha másik változatot
kell rendelnie.

A hordtáskára vonatkozó
utasítások
A FreeStyle Comfort készülékének a hordtáskába
való behelyezéséhez és rögzítéséhez kövesse az alábbi
utasításokat.

540- HU PN MN232-C4 L | Felhasználói kézikönyv

A VÁLLSZÍJ FELHELYEZÉSE
1. A vállszíjat közvetlenül az egységre vagy
a hordtáskára is felhelyezheti.
2. Csatolja a szíjat az egységen vagy a táskán lévőgyűrűre.

FreeStyle ComfortÁrtalmatlanítás és újrafelhasználás A működtetővel közvetlenülérintkező anyagok
Minden esetben juttassa el a FreeStyle Comfort
eszközt az összes alkatrészével együtt az eszköz
forgalmazójához a megfelelő ártalmatlanítás céljából.
Az akkumulátor megfelelő ártalmatlanítására vonatkozó
utasításokkal kapcsolatban a helyi önkormányzattal
is kapcsolatba léphet. Az egység újrafelhasználható
más betegnél, de előtte regenerálás céljából vissza kelljuttatni a berendezés szolgáltatójához.

WEEE és RoHS irányelvek
Ez a szimbólum a berendezés tulajdonosátarra figyelmezteti, hogy a készüléket annak élettartama
végén az elektromos és elektronikus készülékek hulladékainak kezeléséről szóló irányelv (WEEE) szerint
egy újrahasznosító létesítménybe kell juttatnia.

• Koncentrátor burkolata: PC/ABS• Koncentrátor vezérlőpanelje: EBG7 poliészter
vagy ezzel egyenértékű anyag• Oxigénkimenet, csőcsatlakozás: Alumínium
• Egység címkéje: Lexan• Kábelcsatlakozók: Polikarbonát/vinil-klorid• Tápkábel: PVC, fém• Tápegység: Lexan 940 (polikarbonát)• Akkumulátorcsomag: PC/ABS• Akkumulátorcsomag és tápegység címkéi:
Poliészter-film• Koncentrátor hordtáska: Cordura, neoprén,nejlonbélés• Szíj: Polipropilén, acetil, vinil, Cordura, neoprén

A termékeink megfelelnek a veszélyes anyagokkorlátozására (RoHS) vonatkozó irányelvnek.
Legfeljebb nyomokban tartalmazhatnak ólmot vagy
egyéb veszélyes anyagot.
VIGYÁZAT: A helyes ártalmatlanításhoz
vegye fel a kapcsolatot a berendezés
forgalmazójával, vagy a helyi állami
hivatallal az ártalmatlanításra vonatkozó
utasításokért.
VIGYÁZAT: A molekulaszűrőt mindig az összes
országos és helyi környezetvédelmi előírásnak
megfelelően ártalmatlanítsa.

PN MN232-C4 L | Felhasználói kézikönyv HU - 541

FreeStyle Comfort
EMC-tesztÚtmutató és gyártói nyilatkozat – Elektromágneses kibocsátás
A FreeStyle Comfort készülék az alábbiakban megadott elektromágneses környezetben való használatra készült.
A FreeStyle Comfort készülék vásárlójának vagy felhasználójának kell biztosítania, hogy azt ilyen környezetben használják.
Kibocsátási teszt
RF-kibocsátás
CISPR 11
RF-kibocsátás
CISPR 11
Harmonikus kibocsátások
IEC 61000-3-2
Feszültségingadozások/feszültségrezgések
IEC 61000-3-3

Megfelelőség

Elektromágneses környezet – útmutatás

1. csoport

A FreeStyle Comfort csak a belső működéséhez használ RF energiát.
Ezért a rádiófrekvenciás kibocsátása nagyon alacsony, és nem
valószínű, hogy interferenciát okoz a közelében lévő elektronikus
berendezésekben.

B. osztály
A. osztály
Megfelel542- HU PN MN232-C4 L | Felhasználói kézikönyv

A FreeStyle Comfort minden létesítményben használható, beleértve
a lakóépületeket, valamint a lakossági célokat szolgáló épületeket
ellátó alacsony feszültségű, nyilvános hálózatra közvetlenül csatlakozóépületeket is.

FreeStyle ComfortÚtmutató és gyártói nyilatkozat – Elektromágneses zavartűrés

A FreeStyle Comfort eszköz az alábbiakban megadott elektromágneses környezetben való használatra készült.
A FreeStyle Comfort eszköz vásárlójának vagy felhasználójának kell biztosítania, hogy azt ilyen környezetben használják.
Zavartűrési teszt
IEC 60601-es tesztszint Megfelelőségi szint
Elektromágneses környezet – útmutatás
Elektromágneses
A padló fa, beton vagy kerámia járólap
környezet –± 8 kV érintkező
± 8 kV érintkező
legyen. Ha a padló műanyaggal vanútmutatásburkolva, a relatív páratartalom legalább±15 kV levegőn át±15 kV levegőn át
30% legyen.
IEC 61000-4-2
Elektromos gyors
tranziens/burst
IEC 61000-4-4

Túlfeszültség
IEC 61000-4-5±2 kV tápvezetékek
esetében±1 kV bemeneti/kimeneti
vezetékek esetében±1 kV vezeték(ek) ésvezeték(ek) között
±2 kV közös mód váltóáramú vezetékeknél±2 kV tápvezetékek
esetében

A hálózati áram minőségének professzionális kórházi vagy otthoni ápolási környezetre jellemzőnek kell lennie.

Nem alkalmazható±1 kV vezeték(ek) ésvezeték(ek) között

A hálózati áram minőségének
professzionális kórházi vagy otthoniápolási környezetre jellemzőnek kell lennie.

± 1 kV differenciál mód
váltóáramú vezetékeknél± 2 kV közös mód kültéri
kimeneti/bemeneti vezetékeknél
0% UT 0,5 ciklusig (0°,0% UT 0,5 ciklusig (0°,45°, 90°, 135°, 180°,
45°, 90°, 135°, 180°,
225°, 270°, 315°)
225°, 270°, 315°)

Feszültségesések,rövid áramkimaradások és feszültségváltozások a bemeneti
tápvezetékekben
IEC 61000-4-11

0% UT 1 ciklusig (0°)

0% UT 1 ciklusig (0°)

70% UT (30% csökkenés
az UT) értékben 25/30
ciklusig (0°)

70% UT (30% csökkenés
az UT) értékben 25/30
ciklusig (0°)

0% UT 250/300 ciklusig(0°)

0% UT 250/300 ciklusig(0°)

A hálózati áram minőségének
professzionális kórházi vagy otthoni ápolási
környezetre jellemzőnek kell lennie. Ha a
FreeStyle Comfort felhasználója folyamatosműködést kíván hálózati áramkimaradás
esetén is, ajánlott a FreeStyle Comfort
szünetmentes tápegységről vagy
akkumulátorról történő működtetése.

A hálózati frekvenciájú mágnesesmezőknek a tipikus professzionális
kórházi vagy otthoni ápolási környezetre
jellemzőnek kell lennie.
MEGJEGYZÉS: Az UT a váltakozó áramú hálózat feszültsége a tesztszint alkalmazása előtt.

Hálózati frekvenciájú(50/60 Hz) mágneses 3 A/mmező IEC 61000-4-8

3 A/m

MEGJEGYZÉS: Egyes esetekben a FreeStyle Comfort alapállapotba állhat vissza egy jelentősebb elektrosztatikus
kisülést követően, és esetleg teljes ki-/bekapcsolási ciklust kell végezni, hogy visszaálljon a normális működés.
A további részleteket lásd a Hibaelhárítás című fejezetben.

PN MN232-C4 L | Felhasználói kézikönyv HU - 543

FreeStyle ComfortÚtmutató és gyártói nyilatkozat – Elektromágneses zavartűrés

A FreeStyle Comfort eszköz az alábbiakban megadott elektromágneses környezetben való használatra készült. A FreeStyle
Comfort eszköz vásárlójának vagy felhasználójának kell biztosítania, hogy azt ilyen környezetben használják.
Zavartűrési teszt IEC 60601-es tesztszint
Megfelelőségi szint
Elektromágneses környezet – útmutatás
10 Vrms
A hordozható és mobil rádiófrekvenciás
Vezetett RF
10 Vrms
hírközlő berendezéseket tilos közelebb
IEC 61000-4-6
150 kHz - 80 MHz
vinni a FreeStyle Comfort eszközhöz, illetve
annak bármely részéhez (a kábeleket
is beleértve), mint a jeladó működési
frekvenciájára vonatkozó egyenlettel
10 V/m
10 V/mSugárzott RF
kiszámítható ajánlott elkülönítési távolság.
80 MHz-től 2700 MHz-ig
IEC 61000-4-3
Ajánlott elkülönítési távolság
d = 1.2 P

d = 1.2 P = 80 MHz – 800 MHz
d = 1.2 P = 800 MHz – 2700 MHz
ahol P az adó gyártó által megadottlegnagyobb kimeneti teljesítménye wattban(W), és d a javasolt elkülönítési távolság
méterben (m).
A helyhez kötött rádiófrekvenciás adókból
származó térerők, a helyi elektromágneses
felmérés alapján meghatározva,a értékének
az egyes frekvenciatartományokra
vonatkozó megfelelési szint alatt kelllennie. b
A következő szimbólummal ellátott
berendezések közelében interferencia
léphet fel:

1. MEGJEGYZÉS: 80 MHz-en és 800 MHz-en a nagyobb frekvenciatartomány érvényes.
2. MEGJEGYZÉS: Ezek az irányelvek nem minden esetben érvényesek. Az elektromágneses hullámok terjedését befolyásolja az építmények, tárgyak és emberek elnyelő és visszaverő hatása.
a
A helyhez kötött adók, például a rádiótelefonok (mobil/vezeték nélküli) és mobil földi rádiók bázisállomásai, az amatőr
rádiók, az AM és FM rádióadások és a televízióadások elektromágneses térerősségét nem lehet elméleti úton pontosan
meghatározni. A helyhez kötött rádiófrekvenciás adókból származó elektromágneses környezet felmérése érdekébenérdemes lehet helyi elektromágneses felmérést végezni. Ha a mért térerősség meghaladja a vonatkozó rádiófrekvenciás
megfelelési szintet azon a helyen, ahol a FreeStyle Comfort készüléket használják, akkor a FreeStyle Comfort készüléket
figyelni kell, és meg kell győződni a zavartalan működéséről. Ha rendellenes működést tapasztal, akkor további intézkedésekre, például a FreeStyle Comfort készülék elfordítására vagy áthelyezésére lehet szükség.
b

A 150 kHz – 80 MHz frekvenciatartományban az elektromágneses térerősségnek 10 V/m-nél kisebbnek kell lennie.

544- HU PN MN232-C4 L | Felhasználói kézikönyv

FreeStyle Comfort
Ajánlott elkülönítési távolságok a hordozható és mobil rádiófrekvenciás távközlési
eszközök, illetve a FreeStyle Comfort készülék között

A FreeStyle Comfort eszköz olyan elektromágneses környezetben való használatra szolgál, amelyben a sugárzott
rádiófrekvenciás zavarok szabályozva vannak. A FreeStyle Comfort eszköz vásárlója vagy felhasználója segíthet az
elektromágneses interferencia megelőzésében azzal, hogy betartja a hordozható és mobil rádiófrekvenciás kommunikációs
eszközök (jeladók) és a FreeStyle Comfort eszköz közötti, a kommunikációs eszköz maximális kimeneti teljesítményétől
függő ajánlott elkülönítési távolságot.
A jeladó névleges
Elkülönítési távolság a jeladó frekvenciájától függően
maximális kimenetimteljesítménye
150 kHz – 80 MHz
80 MHz – 800 MHz
800 MHz – 2,5 GHz
d = 1.2 P
d = 1.2 P
d = 1.2 P
W
0,01
0.12
0.12
0.23
0,1
0.38
0.38
0.73
1
1.2
1.2
2.3
10
3.8
3.8
7.3
100
12
12
23
Olyan jeladók esetében, amelyek névleges maximális leadott teljesítménye nem szerepel a fenti táblázatban, az ajánlott delkülönítési távolság (méterben) az adó frekvenciájára alkalmazható egyenlettel becsülhető meg, ahol P a jeladó névleges
maximális leadott teljesítménye wattban (W) a jeladó gyártója által megadott adatok szerint.
1. MEGJEGYZÉS: 80 MHz-nél és 800 MHz-nél a nagyobb frekvenciatartományra vonatkozó elkülönítési távolságérvényes.
2. MEGJEGYZÉS: Ezek az irányelvek nem minden esetben érvényesek. Az elektromágneses hullámok terjedését
befolyásolja az építmények, tárgyak és emberek elnyelő és visszaverő hatása.

BesorolásÁramütés elleni védelem típusa:
A II.		érintésvédelmi osztály kettős
szigeteléssel valósul meg.
Áramütés elleni védelem mértéke:	BF típusú berendezés, amely adott szintű
érintésvédelmet		biztosít az alábbi körülmények esetén:		1) megengedhető szivárgóáram;		2) védőföldelés csatlakozásának megbízhatósága
		(ha van).
		Nem szolgál közvetlen kardiológiai alkalmazásra.
A Gyógyászati villamos készülékekre vonatkozó
szabványnak való megfelelés független tesztelése:	A tesztelés az• IEC 60601-1 szabvány 3.1-es kiadásávalösszhangban készült: Gyógyászati villamos
készülékek – 1. rész: Általános biztonsági
követelmények.
• CAN/CSA C22.2 60601-1-8 2.1 számú kiadás,
Gyógyászati villamos készülékek – 1. rész:Általános biztonsági követelmények.

	• A vizsgálatot az RTCA/DO160
	21. szakasz M kategória előírásaival összhangban
végezték.
• CISPR 11 / EN 55011 B osztály 1. csoport,	„Ipari, tudományos és orvosi (ISM)	berendezés”.
• FCC 15. rész, B. alrész – B. osztály	Véletlenszerűen sugárzó berendezések.
IP22 – Ujjak és 12,5 mm-nél nagyobb átmérőjű hasonló tárgyakkal szembeni védelem. A függőlegesen
csepegő víz nem károsítja, ha a burkolat legfeljebb
15°-ban dől meg a normális helyzetéhez képest.
Megengedett tisztítási és fertőzésmegelőzési módszer:
Lásd a jelen FreeStyle kézikönyv „Tisztítás, ápolás
és helyes karbantartás” című részét.
A készülék biztonságosságának foka gyúlékony
altatógázok jelenlétében:	A berendezés nem alkalmas ilyen használatra.
Működési mód: Folyamatos üzemelés.

A berendezés és az egyéb eszközök közötti potenciális
elektromágneses vagy egyéb interferencia elleni
védelem.
	• A vizsgálatot az EN 60601-1-2 /IEC 60601-1-2:
4. kiadású szabvánnyal összhangban végezték.

PN MN232-C4 L | Felhasználói kézikönyv HU - 545

www.ca ire inc .c om

CAIRE Inc.
2200 Airport Industrial Dr., Ste. 500
Ball Ground, GA 30107 U.S.A.
Medical Product Service GmbH
Borngasse 20
35619 Braunfels, Germany
Tel: +49 (0) 6442-962073
E-mail: info@mps-gmbh.eu
Accumed Sagl Viale
Serfontana 106834 Morbio Inferiore
Switzerland
CAIRE and CAIRE Inc. are registered trademarks of CAIRE Inc. Please visit
our website below for a full listing of trademarks. Trademarks: www.caireinc.
com/corporate/trademarks/.
© Copyright 2024 CAIRE Inc. All Rights Reserved. CAIRE Inc. reserves theright to discontinue its products, or change the prices, materials, equipment,
quality, descriptions, specifications and/or processes to its products at any
time without prior notice and with no further obligation or consequence. All
rights not expressly stated herein are reserved by us, as applicable.

|MN232-C4-L[~
10 September 2024

PN MN232-C4 L