SB series

SPEC. SHEET

Thermocouple Transmitter

Model:SBE

Features
Simple wiring using a plug-in socket
Compact
3-port insulation (Input-Output-Power)

Model
SBE --Input
01: K
02: J
03: R
04: S
05: B
06: E
Output
01: 4 to 20mA DC
02: 0 to 20mA DC
03: 0 to 12mA DC
04: 0 to 10mA DC

07: T
08: N
09: PL10: W5Re/W26Re
11: W3Re/W25Re

05: 0 to 1V DC
06: 0 to 5V DC
07: 1 to 5V DC
08: 0 to 10V DC

Power supply
0: 100 to 240V AC
1: 24V DC

How to orderSpecify a model.
(e.g.) SBE-0101-0
0 to 400

Input specifications
Input resistance: 1M or more
External resistance : 100 or less,However, B: 40 or less
Burnout: Upscale, Downscale (Selectable by the DIP switch)
Input:
Thermocouple
Input range
K-200 to 1370
-328 to 2498
J-200 to 1000
-328 to 1832
R-50 to 1760
-58 to 3200
S
-50 to 1760
-58 to 3200
B
0 to 1820
32 to 3308
E-200 to 800-328 to 1472
T-200 to 400-328 to 752
N-200 to 1300
-328 to 2372
PL0 to 1390
32 to 2534
W5Re/W26Re
0 to 2315
32 to 4199
W3Re/W25Re
0 to 231532 to 4199
Minimum span: 50 (100 )Output specifications
When the output range lower limit is zero, (even if zeroadjustment results in a negative value), the output value
will not be negative.
DC current
Allowable
ZeroSpan
Outputload
adjustment
adjustment
range
resistance
range
range
4 to 20mA DC 600 or less0 to 20mA DC 600 or less0 to 12mA DC 1k or less0 to 10mA DC 1k or less
DC voltage
Outputrange
0 to 1V DC
0 to 5V DC
1 to 5V DC
0 to 10V DC-2.5 to 2.5%
0 to 2.5%
0 to 2.5%
0 to 2.5%

97.5 to 102.5%
97.5 to 102.5%
97.5 to 102.5%
97.5 to 102.5%

AllowableZeroload
adjustment
resistancerange
100 or more
0 to 2.5%500 or more
0 to 2.5%500 or more -2.5 to 2.5%
1k or more
0 to 2.5%

Spanadjustment
range
97.5 to 102.5%
97.5 to 102.5%
97.5 to 102.5%
97.5 to 102.5%

Performance
Accuracy• Input: Within 0.2% of each input span, or 2 (4 )whichever is greater
R, S inputs, Less than 200 (400 ):Within 6 (12 )B input, Less than 300 (600 ):Accuracy is not guaranteed.
K, J, E, T, N inputs, Less than 0 (32 ):Within 4 (8 )Cold junction compensation accuracy:
Within 1.0 at -5 to 55
Response time
: 1 sec. (090%)(Average 0.5 sec)
Temperature coefficient: 0.015%/
Insulation resistance : 10M or more, at 500V DC(Input-Output-Power)Dielectric strength: 2.0kV AC for 1 minute(Input-Output-Power)

SBE
Instrument specifications
Case: Flame resistant resin Color: Light gray
Front panel: Polycarbonate
Spring type plug: Polyamide Color: GreenAdjustment: By the front potentiometerZero adjustment : 2.5%
Span adjustment : 2.5%
Indication:
Power indicator (PWR): Green
Lights when the power is turned on.
Flashes in 500ms cycles when an error has occurredin non-volatile IC memory.
Error indicator (ERR): Red
Flashes in 250ms cycles when input is 110% or more.
Flashes in 500ms cycles when input is -10% or less.
Output status selection: Selects Normal or Reverse with
the DIP switch.
No.1 OFF: Normal, ON: Reverse
Burnout: Selects Upscale or Downscale with the DIPswitch.
No.2 OFF: Upscale, ON: Downscale
Momentary power failure: 30msec.
Self diagnosis:
The CPU is monitored by a watchdog timer, and whenan abnormal status is found on the CPU, the unit
restarts with the reset action.

Ambient temperature : -5 to 55
Ambient humidity: 35 to 85%RH (non-condensing)
Weight: Approx. 80g
Mounting method: DIN rail mounting
Be sure to use fastening plates atboth ends of the unit after the unit
is mounted to the DIN rail.
External dimensions : 17.5 (W) x 75 (H) x 85 (D) mm

Environmental specification
RoHS directive complianceFerrules
Made by Phoenix Contact GMBH & CO.
Insulation sleeve attached
ModelConductor cross section
2AI0.25-6BU
0.2 – 0.25mm
2AI0.34-8TQ
0.25 – 0.34mm
2AI0.5-8WH
0.34 – 0.5mm
2AI0.75-8GY
0.5 – 0.75mm
2AI1-8RD
0.75 – 1.0mm
2AI1.5-8BK
1.0 – 1.5mm
2AI2.5-8BU
1.5 – 2.5mm
Crimping pliers
CRIMPFOX ZA3CRIMPFOX UD6Power indicator
Error indicator

Zero adjustment

Circuit configuration, terminal arrangement

Span adjustment+Power supply

Power supply
OutputInput++7-6+5Powercircuit

Outputcircuit

Output+-4+3-2

Input

Input
circuit

CPU1

External dimensions (Scale: mm)
DIP switch
DIN rail

Spring type plug

Installation specifications
Power supply• 100 to 240V AC 50/60Hz
Allowable voltage range: 85 to 264V AC
Power consumption: Approx. 3.5VA• 24V DC
Allowable voltage range: 20 to 28V DC
Power consumption: Approx. 3.5W

SHINKO TECHNOS CO., LTD.

Tel: +81-72-727-6100
URL: http://www.shinko-technos.co.jp/e/E-mail: overseas@shinko-technos.co.jp

20100624