Powering greater
productivity.
Waygate Technologies ISOVOLTTitan|neo X-ray generator

2ISOVOLT Titan|neo generator

Generate better
outcomes.
Empower your operations with theISOVOLT Titan|neo generator, the newest
product in Waygate Technologies
established line of X-ray generators.
Optimized for industry and backed
by excellence, the ISOVOLT Titan|neogenerator powers a range ofradiographic inspection technology.
Beyond being easy to use and maintain,
it delivers reliable, consistent results in
the highest accuracy applications—giving you precision, greater uptime,and peace of mind.

Robust and reliable

Easy to maintain

Backed by excellenceUser-friendly

Highly compatible
3Inspect moreeach day.
Inspect more to achieve more. Get
the reliable, powerful performance
you need to optimize outputs andinspect multiple parts each day,no matter the application.

Performance that never sleeps

The Titan|neo generator offers the best, most consistent
performance available to meet a range of needs, frommulti-cycle exposures to hours of continuous operation.
Its rugged design includes intelligent tube integration andpermanent system monitoring—offering unmatchedramp-up times* and a 100% duty cycle for continuous
operation in inline systems**.

Quality you can quantify

Reduce exposure times for various materials in several
operation modes with excellent dose reproducibility. Thissystem provides high, stable radiation with fluctuations<0.05%. Its extended tube range and maximum current
ensure enhanced imaging contrast and high penetration
power, delivering greater dependability.
*Depending on permissible tube data.
**Subject to optional generator cooling.

4There’s no timefor downtime.
When productivity is a priority,
the ISOVOLT Titan|neo generator
performs. It’s easy to use andmaintain, saving you time, money,and hassle. So you can focus on
optimizing operations.

Convenient control

A user-friendly interface with a modern industrial control
module makes for fail-safe and simple operation.

Stress-free service

A modular design and patented maintenance-freehigh voltage connectors make for easy, non-disruptive
installation, operation, service, and replacement—reducing
the time and resources needed for any repairs, anddecreasing the total cost of ownership.

Safer operations

With a number of built-in safety features and instant,automatic recognition of system status and health, you canenhance operational safety (Performance Level E), maximize
equipment lifetime, and keep your processes under control.

Increased flexibility

Due to the modular structure, it is possible to change fromthe bipolar to the unipolar configuration by simply changing
the connections. This possibility can reduce the need for
additional generators and the related costs for operation
with a wide variety of X-ray tubes and parameters.
Additional functions, such as the automatic adjustment of
the heating current based on the specific high-voltage cable
length, support the simple conversion.

5Tailored to yourspecific needs.
The ISOVOLT Titan|neo generator isdesigned to keep up with increased
pressures and production speeds. Its
enhanced reliability and outstanding
availability improve uptime.
No matter your automation andcustomization requirements, thisgenerator delivers. Its modular,
flexible design, makes for easy
integration into several environments
and external platforms—saving you
time, money, and stress.
When paired with other Waygate
Technologies products, it offers
even greater synergy within yourradiographic inspection system
to reduce unplanned downtime
and maintenance.
The generators are compatible with
Waygate Technologies’ leading
acquisition software Rhythm Insight RT,
so that it can be operated within asystem with a computer and detector– all from a single source.

6

Empower
greater potential.
A broad range of tubes, kits, andaccessories complement integration
and application capabilities—helping
you not just meet your individual
needs, but maximize the potentialof every inspection.
The ISOVOLT Titan|neo is compatible
with the accessories of its previousgenerations, so that it can replace
existing ISOVOLT Titan generators oneto-one without the need for a range ofnew accessories.

Pumps and Coolers

PWL 5000 WTOL 4503

Safety Devices• Primary interlock switch• Alarm box and switch box• Flash and warning lamps• Country-specific safety kits

Voltage Divider

Integration and Solution Kits
• Exposure Calculator• Control via acquisition software Rhythm Insight RT
within a system with a computer and detector• Titan PC (external PC-based visualization)• PROFIBUS extension kit• Optional R24 high-voltage socket instead of themaintenance-free high-voltage socket (in order to beable to continue using R24 cables already in the field)

OW 4002• Optional Water-cooled generator version for use atelevated ambient temperatures

Dosimetry and Calibration Kits
• Voltage divider (incl. calibration certification)

High Voltage Cables• In different standard lengths, with maintenance-free plugs

OLK 507

Technical specifications
High voltage generator

HP160

HP225

Max out voltage kV

160

225

Max out current mA

45

45

Max out power kW

4.5 (limited by tube spec; 1 phase mains:4.0 kW)

4.5 (limited by tube spec; 1 phase mains:4.0 kW)

Insulation

Oil

Oil

Housing dim (w x d x h)

340 x 945 x 750 mm(13.38” x 37.20” x 29.52”)

340 x 945 x 750 mm(13.38” x 37.20” x 29.52”)

Weight195 kg (429.9 lbs)

190 kg (418.87 lbs)

Presel and settings

From 5 to 160 kV in 1 kV

From 5 to 225 kV in 1 kV

Dig display of set and act values

3 digits (set); 4 digits (act)

3 digits (set) ; 4 digits (act)

Display resolution

1 kV (set); 0.1 kV (act)

1 kV (set) ; 0.1 kV (act)

Accuracy<1%<1%

Reproducibility<0.01%<0.01%

Temperature drift<80 ppm/K<100 ppm/K

Presel and settings

From 0.1 to 45 mA in 0.1 mA

From 0.1 to 45 mA in 0.1 mA

Dig display of set and act values

3 digits

3 digits

Display resolution

0.1 mA

0.1 mA

Accuracy<1%<1%

Reproducibility<0.25%<0.25%

Temperature drift<100 ppm/K<100 ppm/K

Programmable timer

11

Presel and setting

1 ... 9999 s

1 ... 9999 s

Dig display of set and act values

4 digits

4 digits

PrewarningAudible and visible

Audible and visible

Presel and setting

2 ... 120 s or deactivated

2 ... 120 s or deactivatedNumber of storable programs

250250Warm-up

Auto mode based on real time clock

Auto mode based on real time clock

X-ray tube set up

8 tube selectable from a databaseof 40 pre-programmed tubes

8 tube selectable from a databaseof 40 pre-programmed tubes

Operation history

Stored on SD

Stored on SD

Warm-up history

Stored on SD

Stored on SD

Tube voltage

Tube current

Exposure time

Programmed mode

8Control module
Dimension wxdxh

440 x 114 x 295 mm (17.32” x 4.48” x 11.61”)

440 x 114 x 295 mm (17.32” x 4.48” x 11.61”)

Weight3.8 kg (8.37 lbs)

3.8 kg (8.37 lbs)

AUX: 1N PE 230 V ±10% 50/60 Hz 10 A,

AUX: 1N PE 230 V ±10% 50/60 Hz 10 A,

MAIN: 3N PE 400/230 V ±10% 50/60 Hz 20 A
or 1N PE 230 V ±10% 50/60 Hz 63 A,

MAIN: 3N PE 400/230 V ±10% 50/60 Hz 20 A
or 1N PE 230 V ±10% 50/60 Hz 63 A,

3-phase, grounded neutral TN-S or TN-C-S
mains (star connected system), optional
3-phase isolation transformer

3-phase, grounded neutral TN-S or TN-C-S
mains (star connected system), optional
3-phase isolation transformer

Separate grounding for X-ray tube and highvoltage generator (minimum 6 mm²)

Separate grounding for X-ray tube and highvoltage generator (minimum 6 mm²)

AUX: 10 A (1N PE)

AUX: 10 A (1N PE)

MAIN: 63 A (1N PE) or 20 A (3N PE)
Time-delay fuses, customer-supplied

MAIN: 63 A (1N PE) or 20 A (3N PE)
Time-delay fuses, customer-supplied

Operating temperature range

0 °C to +40 °C

0 °C to +40 °C

Storage temperature range-30 °C to +70 °C-30 °C to +70 °C

Connected loads

Power connection

Grounding

Mains fuses

9High voltage generator

HP320

HP450HR240

Max out voltage kV320450240

Max out current mA4545

3

Max out power kW

4.5 (limited by tube spec;
1 phase mains: 3.5 kW)

4.5 (limited by tube spec;
1 phase mains: 3.5 kW)

0.320 (limited by tube spec)

Insulation

Oil

Oil

Oil

Housing dim (w x d x h)

340 x 945 x 750 + 340 x 945x 540 mm(13.38” x 37.20” x 29.52”) +13.38” x 37.20” x 21.25”)

340 x 945 x 750 + 340 x 945x 540 mm(13.38” x 37.20” x 29.52”) +13.38” x 37.20” x 21.25”)

340 x 945 x 750 mm(13.38” x 37.20” x 29.52”)

Weight190+140 kg(418.87 + 308.64 lbs)

190+140 kg(418.87 + 308.64 lbs)

170 kg (374.78 lbs)

Presel and settings

From 10 to 320 kV in 1 kV

From 10 to 450 kV in 1 kV

From 5 to 240 kV in 1 kV

Dig display of set and act
values3 digits (set); 4 digits (act)

3 digits (set); 4 digits (act)

3 digitsDisplay resolution

1 kV (set); 0.1 kV (act)

1 kV (set); 0.1 kV (act)

1 kV

Accuracy

<1%

<1%

<1%

Reproducibility<0.01%<0.01%<0.01%

Temperature drift<80 ppm/K<80 ppm/K<80 ppm/K

Presel and settings

From 0.1 to 45 mA in 0.1 mA

From 0.1 to 45 mA in 0.1 mA

From 0.01 to 3 mA in0.001 mA

Dig display of set and act
values3 digits

3 digits

4 digits

Display resolution

0.1 mA

0.1 mA

0.001 mA

Accuracy

<1%

<1%

<1%

Reproducibility<0.25%<0.25%<0.25%

Temperature drift<100 ppm/K<100 ppm/K<100 ppm/K

Programmable timer

11

1

Presel and setting

1 ... 9999 s

1 ... 9999 s

1 ... 32767 s (xs-control)

Dig display of set and actvalues

4 digits

4 digits

5 digits

Prewarning

Audible and visible

Audible and visible

Audible and visible

Presel and setting

2 ... 120 s or deactivated

2 ... 120 s or deactivated

2 ... 255 s or deactivated

Tube voltage

Tube current

Exposure time10

Programmed mode
Number of storable
programs

250

250—Warm-up

Auto mode basedon real time clock

Auto mode basedon real time clock

Automated intelligent
tube conditioning

X-ray tube set up

8 tube selectable froma database of 45
pre-programmed tube

8 tube selectable froma database of 45
pre-programmed tube—

Operation history

Stored on SD

Stored on SD—

Warm-up history

Stored on SD

Stored on SD—

Dimension (w x d x h)

440 x 114 x 295 mm(17.32” x 4.48” x 11.61”)

440 x 114 x 295 mm(17.32” x 4.48” x 11.61”)—

Weight3.8 kg (8.37 lbs)

3.8 kg (8.37 lbs)—

AUX: 1N PE 230 V ±10% 50/60
Hz 10 A,

AUX: 1N PE 230 V ±10% 50/60
Hz 10 A,

MAIN: 3N PE 400/230 V ±10%50/60 Hz 20 A or 1N PE 230 V±10% 50/60 Hz 63 A,

MAIN: 3N PE 400/230 V ±10%50/60 Hz 20 A or 1N PE 230 V±10% 50/60 Hz 63 A,

3-phase, grounded neutral
TN-S or TN-C-S mains(star connected system),
optional 3-phase isolationtransformer

3-phase, grounded neutral
TN-S or TN-C-S mains(star connected system),
optional 3-phase isolationtransformer

Control module

Connected loads

Power connection

Grounding

1N PE 230 V ± 10% 50/60 HZ
10 A AUX,
1N PE 230 V ± 10% 50/60 HZ
10 A MAIN

Separate grounding for
Separate grounding for
Separate grounding for
X-ray tube and high voltage X-ray tube and high voltage X-ray tube and high voltage
generator (min. 6 mm2)
generator (min. 6 mm2)
generator (min. 6 mm2)
AUX: 10 A (1N PE)

AUX: 10 A (1N PE)

MAIN: 63 A (1N PE) or 20 A
(3N PE) time-delay fuses,
customer-supplied

MAIN: 63 A (1N PE) or 20 A
(3N PE) time-delay fuses,
customer-supplied

Operating temperature
range

0 °C to +40 °C

0 °C to +40 °C

0 °C to +40 °C

Storage temperature range-30 °C to +70 °C-30 °C to +70 °C-30 °C to +70 °C

Mains fuses

10 A (1N PE) integrated
into aux switch,
10 A (1N PE) integrated
into main switch

11

A legacy of excellence.
A future of possibility.
With more than 125 years of X-ray experience and thousands of installations
worldwide, the ISOVOLT platform is known for its reliability and performancein the radiography and NDT space. It provides a standard for X-ray
generators everywhere. The ISOVOLT Titan|neo generator is just one partof Waygate Technologies revolutionary X-ray, CT technology, and digitalradiography suite, and just one way Baker Hughes is setting the stage for
future innovations designed to improve your operations.

Waygate Technologies, a Baker Hughes business
Bogenstr. 4122926 Ahrensburg
Germany
Tel.: +49 4102 807 0Fax: +49 4102 807 277
Waygate Technologies, a Baker Hughes business
201 Beltway Green Blvd.
Pasadena, Texas 77503
Tel.: +1 281 542 3600

bakerhughesds.com/waygate-technologies

Copyright 2021 Baker Hughes Company. All rights reserved.
BHFF30228A (03/2023)

waygate-tech.com