QxL & QxPAdvanced Rasterizer and Waveform Monitor for
Hybrid IP/SDI, 4K/UHD, HDR/WCG
Generation, Analysis and Monitoring1

Qx Series - Technology to power change
From the moment you first power up a Qx Series instrument, you’ll appreciate
the attention to detail in a platform designed to meet the increasing demands ofmonitoring and testing in SDI/IP hybrid environments. The Qx Series is equally athome in master control rooms, OB and link trucks, production studios, technical
QC, product development, engineering compliance testing and operational systemmonitoring. Whether you are working in SD, HD or UHD, SDR or HDR, SDI or IP,
conventional or remote production, Qx rasterizers and waveform monitors bring
together the user-configurability and advanced tools required for full operational
flexibility when transitioning to your next generation workflows.
Available in three platforms, with a common look and feel, the Qx Series provides anintuitive user interface and toolsets that help with rapid fault diagnosis and reduce
the need for staff training. The comprehensive feature set supports
SD/HD/3G/6G/12G-SDI, 10G/25G IP interfaces, and SD/HD/UHD, IP SMPTE 2022-6,
SMPTE 2110-10/20/30/31/40 (ST 2110-20 RGB payloads up to 21Gbps) with ST 2022-7,
PCAP, Dolby E Decode and AMWA NMOS, easing system design and future-proofing
your investment.

Analyzer/Generator - Simultaneous operation
The QxL and QxP provide simultaneous Generation and Analysis for a wide rangeof ST 2110-20/30/31/40, 2022-6 and SD*/HD/3G/6G/12G-SDI formats with support for
up to 80 channels of 48 kHz Class C audio in 2110-30/31 and up to 128 channels over
12G SDI.
Configuration can be either Manual, or under REST API control enabling automatedclosed-loop testing for Engineering regression and manufacturing. ST 2110
Generation and Analysis is also NMOS enabled for ease of integration into IP based
systems.
With a full suite of SDR Rec. BT 709/2020, plus native and mapped Wide Colour
Gamut (WCG) HDR patterns in HLG, PQ, S-Log3 and SR-Live formats, you areequipped for flexible broadcast SDR and HDR operation.
Generator and Analyzer video format, colorimetry and transfer function can all be
configured independently. You have the flexibility to send out an EUHD Rec BT.2100
HDR 12 bit RGB pattern with Class C audio and PTP locked Timecode and analyze
the down-converted, down-mapped HD SDR Rec 709 return simultaneously.

Compliance - it’s all about Test and Measurement
Developing products or commissioning the latest equipment is more than just implementation. Equipment has to be tested
against the required standards for it to be considered fit for purpose.
In the 12G-SDI world, noise floors are required to be much lower to ensure that
accurate and meaningful measurements can be taken. Qx SDI generation andmeasurement technology has been specifically adapted for 12G applications. With its
unique class leading SDI-STRESS toolset, sophisticated RTE™ (Real-Time Eye) multirate physical layer display, and automated SMPTE compliance measurements, theQx Series offers a single product solution for SDI compliance verification.

If you are working in SMPTE ST 2110, with ST 2059 Precision Time Protocol (PTP),a core IP toolset, available in both the QxL and QxP offers an operator all of the IP
confidence status monitoring in an intuitive and accessible manner. The optional
IP-MEAS test suite provides a comprehensive set of tools for compliance verificationand commissioning of your IP systems and equipment.
Hardware-based timestamping locked to PTP ensures accurate, realtime,
deterministic timing measurements of media flows and ST 2110-21 buffer models.

2

QxL / QxP

* Upcoming software releaseApplications

Outside Broadcast
NEP UK selected Qx rasterizers for two of its new OB trucks,
for use at major events and sporting fixtures. Hybrid SDI/IP
capability was a key selling point for NEP enabling them toaccommodate clients whether they are using conventional
SDI or have made the move to IP. The ease of use of the Qx
was also a major factor, making it quick and simple for both
NEP engineering staff and freelancers to learn and use.
Engineering and Technical Director, NEP UK, said, “We’ve
been very happy with the reliability of PHABRIX test andmeasurement equipment in the past, so it was an obvious fit
to look at the Qx for these new IP-capable vehicles.”Sports and Live Events
PHABRIX recently concluded nine months of successful HDR
technology trials with BT Sport in the run-up to the launch of
BT Sport Ultimate. The Qx is now deployed to monitor andanalyze SDR and HDR Wide Color Gamut (WCG) material on
the live system. PHABRIX supported BT Sport, providing its
Qx rasterizers and technical expertize, as they developed andrefined their live production workflow for the launch of their
new HDR, UHD and Dolby Atmos® supported proposition. On
the bench PHABRIX collaborated with BT Sport to analyze
and provide suggested settings for SDR to HDR convertersand ‘tone mappers’ used in the trucks and throughout the
network.

Manufacturing & Compliance Testing
Mellanox Rivermax® development and regression testingteams have been using the QxL to provide simultaneous analysis of the SMPTE ST 2110 Video, Audio and ANC DATA flows
from their Rivermax® video streaming library for media andentertainment, running on Mellanox ConectX-5 and newer,
Network Interface Cards,” said Nir Nitzani, senior director SW
development at Mellanox Technologies.
“The ability to install the QxL in the machine room andremotely access and control the realtime measurementsfrom several sites has been an ideal fit with our engineeringdevelopment workflow."

Extended Reality (XR)
7thSense chose a PHABRIX QxL 25G IP ST 2110 rasterizer for
SDI and advanced IP 2110 product verification in-house, andat onsite installations. PHABRIX and 7thSense joined forces
to develop the capability to output the next generation of ST
2110 IP formats, including UHD/4K 12 bit 444 60p.
Richard Brown, CTO, 7thSense, said, “As we begin delivering
SMPTE 2110 support from our Delta Media Server and Juggler
pixel processor products, we needed to ensure all of oursolutions comply to the required specifications. We needed
test and measurement technology that we could rely on, wasrobust, and supported the wide variety of formats we need totest."

www.leaderphabrix.com

3

Platforms to suit every workflow
The flexible architecture of the QxL and QxP offers many engineering grade tools as standard, together with in-field optional
upgrades for UHD/4K, 2110-20 EUHD (RGB UHD/4K 48-60p), PCAP, Dolby E Decode, HDR, and Audio/Video/ANC test signal
generation. A factory-fitted hardware option provides RTE™ realtime SDI eye and jitter analysis with the further option of a highly
advanced SDI-STRESS toolset.

PHABRIX QxL - 10/25GbE / 12G-SDI

For realtime UHD IP workflows on 25G networks with video payloads up to 21 Gbps, the class-leading QxLprovides support for ST 2110 and 2022-6 on generic 10G/25G SFP28 interfaces. The QxL is 10G IP-enabled
as standard, with support for simultaneous generation and analysis of a JT-NM TR 1001-1:2020, ST 211020 (video), up to four 2110-30 (PCM) and 2110-31 (AES transport) audio and a 2110-40 ANC media flow, allwith 2022-7 Seamless IP Protection Switching (SIPS) and AMWA NMOS IS-04 discovery and IS-05 device
connection management.
Independent PTP followers on both media ports are provided for fully-redundant media network operation
with AMWA NMOS IS-04 discovery and IS-05 device connection management. The option of HDR, PCAP,
Dolby E Decode and IP-MEAS in-field license upgrades means that you can tailor your system to your current
needs while retaining full flexibility for the future.
Support for 25G IP, UHD/4K formats for both IP and SDI, including some HD/2K extended mode formats,
PCAP, IP Measure, and UHD 2110 Extended Mode formats (YCbCr/RGB 444, 8-/10-/12-bit; 48 to 60 Hz), canalso be added as optional licenses (for the full list of UHD and EUHD standards supported, please see pages
26 - 28).
SDI BNC and SFP media interfaces are available as a factory-fitted option. The SDI Eye and Jitter hardware
option and the unique SDI-STRESS toolset provide all the tools for SDI physical layer analysis andcompliance testing.

PHABRIX QxP - 10/25GbE / 12G-SDI
KeyStandard
Optional

Introducing the latest member of the Qx test and measurement family - the QxP, the world’s first portable, 12G-SDI, 25G ST 2110,
combined waveform monitor, generator and analyzer, with mains and external DC power and a choice of V-mount or Gold mount(G-mount) external camera battery plate. This provides all of the functionality of the QxL in a handy, lightweight, portable 3RU
chassis with an integral 1920 x 1200 7 inch LCD multi-touch screen. If you prefer buttons or mouse control then you are free to
use any combinations of controls.
You can run all QxP instruments on the integral screen with minimal retraining. Plug in an external HDMI monitor and you havethe same experience as if you were using the QxP as a conventional Rasterizer.

4

QxL / QxP†JT-NM Tested - For more details on the JT-NM Tested Program (Spring 2020) and its test results, please see:
http://jt-nm.org/jt-nm_tested/Simplicity - an interface that puts you in control
The Qx’s innovative app style interface is a radical change from traditional test and measurement systems. Intuitive mouse control
with context-driven dropdown menus hides the complexity of modern SDI and IP systems providing an uncluttered view of criticalinformation. Instruments can be resized, the system auto-presenting more information as the screen area permits.
The Qx offers a fully flexible user-defined instrument layout, displaying up to 16 instruments on a single 1920 x 1080 display.
Individual instruments can switch between 1/16th, 1/4 or full screen. With an output frame rate of 50, 50.94 or 60 Hz to match thevideo format, the GUI has adjustable brightness for controlled lighting environments.

Instrument Tooltips• The UI displays context-driven

tooltips providing additional
information when hovering overparameters

Screen Capture• SFTP and Browser network access to event
logs, screenshots, user presets

Instrument Navigation• Each instrument includes a pop-up submenu,
giving you access to the configuration
parameters of that instrument

Presets• Save multiple display layouts as presets andupdate as required• Save bespoke layouts for different
operational tasks• Use to change rapidly between different

Network Time Protocol• Configure the unit to use a date and timetransmitted by a target Network Time
Protocol (NTP) server or set the time anddate manually in the Time and Date dialog.

screen layouts eg. Audio, HDR or IP focus

Instrument Launch Menu

• Provides access to the instruments andother system menus available on the unit

• Each Instrument available in the menu islisted alongside a designated icon

Numeric Slider & Scroll Bar• Adjust numeric values by dragging orError Highlighting• Errors are displayed in red fontscrolling the slider button• Mouse over the numeric field and scroll forfine control• Connect to USB keyboard, click and enter
specific alpha numeric values

www.leaderphabrix.com

5

QxP Touchscreen - control at your fingertips
Instrument Area

• Configure and display up to

16 layouts with up to 16 1/16
screensize instruments visibleon each layout• Touch-sensitive Controls• Open Each Instrument at
Default 1/4 Screen Size

• Tooltips at a touch

Context-sensitivemenus

7 inch LCD Multitouch
Display• Instrument-specific,

Context-sensitive Menus• Easy Access to

Configurable Parameters

Rotary Control• Move Cursor Focus Between
Instruments• Navigate & Select fromContext-sensitive Menus

On/Off• Press for OK

Speaker

PHABRIX Key• Display Instrument Launch

CursorControl
Buttons

Menu• Press Repeatedly to Returnto Default Function Bar

PHABRIX Button• Display Instrument Launch

2x USB Ports• Press Repeatedly to Return

Headphone
SocketMenu

to Default Function Bar

Function Bar• 10 Context-specific Function
Softkeys• Access Configuration Menus
of Active Instruments

Submenus
Function Buttons• 10 Physical Function Button
Controls• Correspond Directly to theContext-specific Function
Softkeys Above6

QxL / QxP• Frequently-used Button Banks• Rapid Access to Favorites,
Presets & System

Introducing our new waveforms
PHABRIX is pleased to announce the development of class-leading Waveform Monitoring for the Qx Series of Rasterizers andWaveform Monitors. Utilizing a technique patented by PHABRIX to efficiently deliver a high-resolution image processing pipeline
with support for deep color sources up to 12-bits, this instrumentation delivers all the fine detail required for Camera Shading,
Image Grading or critical QC of both SDR and HDR content.
A choice of Overlay, Stacked and Parade display modes are provided each with the option of multi-colored, highlighted, green
or monochrome traces. The flexibility to display YCbCr, RGB, YRGB, YGRB and individual components is retained along withconnected instrument cursor linked to Picture and Data view, and user markers linked to Vectorscope. Single Line Mode and H
and V magnification are available for detailed inspection.
Luminance Nits scales and operation user-controlled Nits markers are provided for SDR, HLG, PQ, S-Log3, SR-live HDR formats.
Both SMPTE-narrow and full-range operation are supported along with matrices for 709, 2020 and DCI P3 over the wide-range
of YCbCr:422, RGB:444, SDI, 2110, SD*/HD/2K/UHD/4K/EUHD formats for which PHABRIX is famous.

New Overlay Mode
Accurate reproductionof image fine detailwhile simultanously
retaining bright
instrument highlights

Full-width waveform
display picture Modeto match Picture width

Overlay RGB (Color)

Stacked YRGB (Color)

Parade YCbCr (Color)

YRGB Parade (Color Highlight)

YRGB Parade (Green)

YCbCr Parade (Monochrome)www.leaderphabrix.com

7

Standard Toolset
As standard, the Qx Series offers a flexible user-defined instrument layout displaying up to 16 simultaneous windows, and theability to change rapidly between bespoke layouts for different operational tasks with user presets.
Picture view, waveform monitor, vectorscope, 32 channel audio metering, decoded audio channel status information, detection ofcommon Dolby formats, ANC status and payload, on screen display of OP47 and CEA-608 in 708 closed captions and Ancillary
Time Code (ATC), Loudness monitoring, and advanced control and logging with human readable event logs, remote operator GUI
access over noVNC and a full REST API are all provided as standard.

Picture Display

CEA-608 Sous-titres Codes
Decoder et afficher

01:34:27:02

01:34:27:02

Auxiliary Data Decode

Analyzer - Picture Copy

• Cursors linked to Waveform and Data View
• Action, graphics and user-definable custom• Closed Captions OP47, CEA-608 in 708• Primary Closed Caption decode picture• Secondary closed caption decode: Monitor

window• ANC Timecode with OSD• Date, V-chip, AFD and Input name
• SCTE 104 indication and logging

608/708 closed captions in a second
language, or compare different screen safeareas
• Independently manage overlay elements
including; Closed Captions, Picture Safe
Areas, V-chip, AFD, SCTE 104, Image Center
Crosshairs

Analyzer - Waveform

Analyzer - Vectorscope

Analyzer - Audio Meters

Rec. 2020 and HDR formats• Custom 'user markers' linked to Waveform• Center on target or custom user markers• 0.5x to 4x Mag, center on chosen target• Single line mode linked to Picture Cursor• Tooltip display of Cb, Cr and Hue Angle• IQ axis on/off• 12-bit processing

monitoring a block of up to 16 channels at a time,
for a total of up to 32 channels of audio metering• 2110 audio group display across up to 4 flows• Ballistics: PPM-I, PPM-II, Vu, Vu-Fr, Fast
• Scales: dBFS, dBu -18, dBu -20, BBC, DIN45406,
NordicN9• Adjustable peak hold times: Off, 0.1 s to Inf• Audio pair correlation meters, numerical level• Detection of Dolby E, ED2, DD, DD+, DE line pos• Stereo/mono audio preview bus

safe areas• 1/16, 1/4 or full size display• YCbCr, YGBR and GBR display modes• Cursor linked to Picture and Data View
• Single line mode linked to Picture Cursor• Configurable H and V Graticules• User markers• Overlay, Stacked, Parade, Single line, H& V Mag, Brightness, Persistence andMonochrome controls• 12-bit processing• Horizontal or Vertical Measurement cursors

Audio Status• 32 channel indication of audio type andpresence, PCM, Dolby E, DD, DD+, ED2• Decoded channel status information for up
to 128 channels• Clear indication of useful audio parameters
including CRC, PCM/data, sample
frequency, word length• Channel Status data view (Hex)8

CEA-608 Closed Caption
Decode and Display

QxL / QxP

• 75% and 100% Targets for ITU-R Rec. 709,

• Two meter windows can be opened, each

Loudness Monitoring

Analyzer - Ancillary Status

short term and integrated loudness• User control of integrated, momentary andshort term targets• User-adjustable true peak alarm threshold• Loudness logging stored automatically• Grid View – clear visual overview, present/• EBU R128 and ITU-R BS.1770• Indicators for true peak, range, momentary,• SMPTE ST 291 VANC/HANC ancillary data
presence/status window
absent/fault indication• List View – ANC present list with location andstatus information for Checksum, Parity, DBN• Link to ANC Inspector• Tooltip provides ST 291 ANC type overview

Standard Toolset

Data View Analyzer with ANC Inspector

The engineering grade Data View Analyzer and ANC Inspector tools provide easy, accessible visualization of the data on
an SDI interface and associated ANC packets. Deep SDI data inspection with full freedom to inspect Active Picture, VANC,
HANC and API controls to read back Active Picture Data under automation control is included. Also featured is ANC packet
decapsulation and error reporting for detailed analysis and debug of ANC payloads.

Analyzer - Data View

• Allows analysis of complex faults• Detailed view of data words in the SDI
stream with tooltip hint

• Navigate function for rapid access to arequired line, pixel or TRS word

• Color-coding to help identification• Cursor linked to Picture and Waveform

ANC Inspector• Ancillary data packet analyzer• Link from ANC Status window• User-defined DID/SDID windowed search• Trigger on error, single shot, continuous• ANC packet capture with Hex view
• ANC packet decode view

I/O and Reference Configuration

Video Timing & System
Reference (2022-6/SDI†)• Measurement of the timing of inputs againstreference• Indication of reference status and stability• Indication of the relative co-timing of input
SDI channels• Graphical and numeric display

System IO• Shows the status of signal inputs & outputs,
external reference, cable length, andconnector details• View and switch analyzer input using
connector icons (IP or SDI)

• SDI: Select BNC or SFP I/O, cable type, loop
through and generator copy outputs• IP: Active IP SFP receive inputs and transmit

AES IO Config• Four versatile bi-directional AES unbalanced
interfaces• Audio meter monitoring pair, generator
audio output or AES input• SDI Input to AES Output de-embedder forboth PCM and Dolby encoded audio• Route AES Input signals to other AES

outputs providing up to three copy outputs

outputs are indicatedAnalyzer - Video Standard CRC Analysis (2022-6/SDI†)(2022-6/SDI†)• Display of detected SMPTE S352 Payload
ID for each SDI Link and Subframe• Manual override of S352 ID• Selection of SMPTE video format• Indication of S352 errors• Check for CRC errors on Y, C and ANC• Reporting of the number of SDI input

failures, the last failure time, total analysis
time and error rates• Detect active picture changes and view theactive picture CRC to observe any changes
in the expected active picture CRC value

Stats - SDI In (2022-6/SDI†)• Cable length indication• Indication of data rate and clock divisor• Reporting of active and total pixel and line
counts• Y and C payload ID• SDI switch line CRC masking control, for
SMPTE RP168 compliance checking† Available with SDI and PHQXL01-3G / PHQXP01-3G or PHQXL01E-3G / PHQXP01E-3G

www.leaderphabrix.com

9Standard Toolset

ST 2110 and ST 2022-6 Monitoring
The core IP feature set, provided as standard in the QxL and QxP, offers an operator all of the ST 2110 confidence statusmonitoring in an intuitive and accessible manner.
The toolset supports simultaneous decapsulation of one video, four audio and one ANC Data flows. Supported SMPTE protocols
include ST 2059 (PTP) ST 2110-20 (Uncompressed Video), -30 (PCM Digital Audio), -31 (AES3 Transparent Transport) and -40(ANC Data). ST 2022-7 seamless protection (SIPS) with AMWA NMOS IS-04, IS-05 and PTP system resource, is provided overtwo media network interfaces using industry standard optical ethernet SFPs. Audio handling conforms to ST 2110-30 Class C withsupport for 48 kHz streams from 1 to 10 channels at packet times of 1 ms and 1 to 80 channels at packet times of 125 μs.
Also provides an indication of the timing relationship of each of the eight ST 2022-7 flows to PTP with status information, as
well as a ST 2022-7 status tool that reports the health and relative timing skew of each ST 2022-7 pair, all with hardware timestamping.

SMPTE 2110 & 2022-6

SFP IP Network• Reporting of presence of SFPs, SFP MAC

and IP addresses (flow source IP address),and interface status• Tx and Rx packet counters for indication of
traffic activity• User configuration of SFP IP Addresses,
Masks, Gateway and DNS addresses

SFP Information

IP Receive Flows

physical network connection• Indication of SFP vendor and laser
characteristics• RX and TX power for debug of fiberconnectivity

receiver and user selection of the requiredflows• Indication of Qx locked status, Protocol, Src
and Dst IP and Port Numbers, SSRC, Packet
Counts, Sequence, payload and CRC errors• Configuration of Multicast Destination IP
addresses and subsequent Multicast Joinrequests• SFP status information for monitoring the

SMPTE 2110 & 2022-6

ST 2022-7 Status

ST 2110 PTP Info - 2 port

• Indication of the health of ST 2022-7• Warning of ST 2022-7 flow-pair mismatch• Warnings of errors on flows and errors on• Indication of lock status• Grandmaster information including leader ID

reconstructed output and error rates per
second• Relative measure of Path Differential offlows on SFPB (Blue Network) relative to
SFPA (Amber Network)• Class A, B,C, D markers

10

SMPTE 2110

• Control of PTP domain and communication

seamless protection

QxL / QxP

• Reporting of the IP Flows available to themode (multicast, hybrid w/o negotiation)and time source• Indication of estimated frequency andphase lock offsets• Indication of one step or two step traffic• Two independent PTP followers

IP Flow Latency• Indication of media latency• Indication of relative timing of audio and
ANC flows wrt video• Indication of relationship of underlying
media to PTP• External analog reference timing wrt PTP

Standard Toolset
AMWA NMOS
A suite of AMWA NMOS tools provides flexibility when integrating with an NMOS controller and associated network topology.
Supported protocols: IS-04 v1.0, 1.1, 1.2, 1.3 IS-05 1.02, 1.1 and IS-09 PTP domain. Provision of both in-band and out-of-band control
topologies with manual, mDNS, DNS-SD and DHCP. Configure Senders and Receivers independently as single or dual NMOS end
points. NMOS troubleshooting is aided by the simultaneous views of the state of both the Sender and Receiver Master and RTP
Enables, SDP, and the IS-05 parameters. The receiver auto-detected video format and audio packet time and channel count arecompared with the received SDP information for diagnosis of the format information supplied by the SDP record.

NMOS Receiver Status

NMOS Receiver SDP

NMOS Receiver IS-05

receiver Master Enable, RTP Enables and
SDP records for each media interface• Available in 1/16 view - toggles with the SDP
view• Display of the Master, RTP and SDP of all
Receiver flows

improved readability• Display adapts with NMOS Dual or Single
receiver configuration (Dual shown)• Individual tabs display IS-05 parameters for• At a glance overview of the state of the• Display of the active receiver SDP record• User-configurable color highlighting for• Display of the active receiver IS-05
parameters

each receiver flow

• Human readable tree view of the IS-05 JSONwith expand/collapse for rapid navigation• Display adapts with NMOS Dual or Single
receiver configuration (Dual shown)

NMOS Sender Status• At a glance overview of the state of theSender Master Enable, RTP Enables and
SDP records for each media interface• Available in 1/16 view - toggles with the SDP
view• Display of the Master, RTP and SDP of all
Generator flows• Display of the Master, RTP and SDP status of
all monitor GUI Interface flows

NMOS Sender SDP• Display of the active sender SDP record• User-configurable color highlighting for

improved readability• Display adapts with NMOS Dual or Single
receiver configuration (Single shown)

NMOS Setup

2110 Format Setup• Manual, mDNS or DNS-SD discovery of the• At a glance comparison of auto-detected,

Registry with DHCP
• Status reporting of registration and DNS
domain• Independent configuration of sender andreceiver as single or dual NMOS endpoints• NMOS node Enable/Disable• IS-09 PTP Domain Enable/Disable• Selectable NMOS endpoints

NMOS Sender IS-05• Display of the active sender IS-05
parameters• Individual tabs for the display of the IS-05
parameters for each generator and GUI
sender flows• Human readable tree view of the IS-05
JSON with expand/collapse for rapid
navigation• Display adapts with NMOS Dual or Single
sender configuration (Single shown)

SDP and manual format settings• User-configurable video format parameters
for ST 2110-20 flows• User-configurable audio format parameters
for ST 2110-30/-31 flows includes packet
time and channel count• Automatic detection of audio format,
channel count and packet timewww.leaderphabrix.com

11Remote Access
Various methods are provided to enable you to establish a remote connection with your QxL or QxP system, depending on yourrequirements.

noVNC• Browser remote access using noVNC

technology to deliver 16 simultaneous
scalable instruments over a remote network

LLDP• Identify port and device to which the QxL/QxP IP interfaces are connected• Restrict information communicated over
LLDP for IT security purposes• Available in both ST 2110 and ST 2022-6

Remote Connectivity

REST API

Event Logger

automated manufacturer testing• SDI Input standard/status• SDI physical layer timing and alignment jitter• Rest API requests• IP-Tx, IP-Rx, Flow and SFP records• Reference Locking• Audio input presence

Mgmt Interface Config

USB File Manager• QxL/QxP can be controlled remotely over a
network via a REST API• Integrated control, monitoring and• Manual or Dynamic Addressing modes• mDNS and DNS• Select Default Gateway from Media orManagement interfaces• Copy presets, instrument logs,

screenshots and user TIFF images to andfrom USB memory stick• Delete selected files• Control access to REST API and VNC• File Transfer: FTP or Browser access toscreenshots and PCAPs, User Test Patterns(TIFF), log files• Remote Software Product Updates• HDMI: UI video (1080p), UI audio (2-ch), local
mouse• SDI: UI video (1080p), UI audio (2-ch), local
mouse• noVNC: UI video (1080p low frame rate),
remote mouse with screenshots• KVM: HDMI or DVI (1080p compressed),remote mouse with screenshots• ST 2110: UI (-20), Audio 2-ch (-30)• UI audio available as analog on D26 (rearpanel)• Machine Control via JSON API• Many KVM Options available - including

Long Distance Connectivity, Cloud-based
solutions, multiple access

12

QxL / QxPOptional Toolsets
Fast, automated 12G-SDI physical layer analysis[PHQXL01E-3G / PHQXP01E-3G]

The Physical Layer Toolset is a factory-fitted option for fast 12G/6G/3G/HD/SD*-SDI† physical layer commissioning, testing anddevelopment. Its RTE™ (Real-Time Eye) Technology instantly highlights any SMPTE compliance issues and its realtime SDI jitter
window provides simultaneous monitoring across five specified frequency bands, jitter histogram and video trigger options.
Built-in automation control allows testing to be performed faster, more reliably and at lower cost. Included in the option are afull range of SDI eye measurements including amplitude, DC offset, transition times, overshoot and health indication with both
amplitude and time histograms, as well as choice of color, heat-map overlays and infinite persistence display.

SDI EYE Analysis

SDI Jitter Analysis• RTE™ (Real-Time Eye) for testing SMPTE compliance with indication of DC offset• Automatic measurements of: DC level, amplitude, rise and fall time, rise/fall overshoot, visual• Realtime SMPTE jitter measurements down

rise time indication• Amplitude and time histograms• Single or multiple eyes with choice of color, heat-map overlay and infinite persistence• Timing and Alignment jitter thermometers• User-definable time measurement cursors

Standard Eye Analysis

Realtime Update of all
Measurements

to 10 Hz• 10 Hz, 100 Hz, 1 kHz, 10 kHz, 100 kHz filters• H, 2H, F, V Trigger• Persistence control none to infinite• +/− 0.25 to +/− 64 UI vertical scale
adjustment• Jitter amplitude histogram

Alignment Jitter
Thermometer UIUser-selectable
Trace Colors(including Heatmap)

User-control of horizontal
timebase (1 to 20 eyes)
Display High-pass Eye& Jitter Bandwidth
Filter

Min/Max Scale and
DC Offset

Amplitude Markers(80% and 20%)

Rise/fall Cross:
Indication of
Measured Transitions

Vertical Histogram
Amplitude
Measurement

Automatic Rise/
Fall Overshoot
Percentage
DC Offset
Transition
HistogramTransition Times &Difference Between
Transition Times† Note: Optional UHD SDI formats require PHQXO-UHD

www.leaderphabrix.com

13

Optional Toolsets
SDI-STRESS Testing[PHQXLO-SDI-STRESS Requires PHQXL01E-3G / PHQXPO-SDI-STRESS Requires PHQXP01E-3G]
The advanced SDI-STRESS option is available for stress testing and R&D evaluations of SDI interfaces up to 12G. A
comprehensive API is included for rapid automation testing. The option includes the ability, under automation control, to insert
SDI clock jitter from 10 Hz to 10 MHz (128 UI max) peak-to-peak, mute any of the SDI outputs, and control the SDI scrambler,
sync-bit insertion, pre-emphasis, rise time and driver amplitude. The SDI-STRESS Eye amplitude measurement provides both
Shorth Mean or Mode, with a histogram overlay and a user-defined window for the exploration of eye amplitude. PseudoRandom Binary Sequence (PRBS) generation and analysis of PRBS-7, -9, -15, -23, -31 allows for deterministic measurement oflink Bit Error Rates (BER).

Adv. Generator Tools

PRBS Analyzer

Pathological Detector• Control of jitter insertion frequency and• Indication of PRBS cumulative received data

• Generator status indication of rate at which

amplitude• SDI scrambler and sync bit Insertion on/off• SDI Bit Error (BER) insertion tool
• Control of SDI driver amplitude +/− 15%• Control of pre-emphasis, rise/fall timeand PRBS type
• Generation of PRBS-7, -9, -15, -23, -31• Reported cumulative errors• Calculated Bit Error Rate (BER)

the video pattern generator is creating SDI
pathological conditions• Indication of PLL and EQ pathological rates
per second• Detection on each active SDI link
• Realtime GPI outputs of pathological detectfor external equipment triggering

Advanced Eye Analysis(Additional features with SDI-STRESS option)User-adjustable
Measurement Window

Eye Time Cursor (Two)User-adjustablewindowedhistogram (blue)

Automatic Amplitude
Measure with Modeor Shorth Mean

Eye Time Cursor
Readings & Time Delta
Eye Time Cursor (One)Measurements in Blue
Taken from User-adjustable
Measurement Window

14

QxL / QxPAutomatic Measurement ofTransition Times and Transition
Time Delta

Optional Toolsets

GEN

Audio and Video Generation[Requires PHQXL0-GEN / PHQX0-GEN]

Simultaneously generate and analyze a comprehensive set of SDI and IP formats with the audio and video generation option.
Moving test patterns with up to 32 channels of embedded audio per link or sub-field (up to 128 channels on 12G interfaces)
is included. The Generator toolset option provides not only the core full screen SDI Pathological stress patterns (Eq, PLL, Clk,
CheckField), but uniquely also allows the user to define a percentage combination of the SDI pathological and conventional
generator patterns up to full frame. Importing TIFF files for checking of HDR/WCG graphics or display and evaluation with usercreated test images is also included. The QxL and QxP offer a ST 2110-20 2K/HD, 4K/UHD video flow generator, 2110-30/-31
80 channel audio generator and 2110-40 ANC flow generator. Uniquely, the QxL and QxP can also generate both pattern and
UI 2022-7 flow pairs. The GUI as a flow offers 1 x ST 2110-20 user interface video and 1 x 2110-30/-31 2.0 stereo monitoring bus
audio with ST 2022-7. An IP Transmit configuration tool provides an at-a-glance view of transmitted flow status and selected
formats.

SDI† Video Generation• Confirms generated Video Standard andTest Pattern details• BNC output, SFP output and sub-image/fullimage mapping information• Video Reference, output offset adjustment
and Jitter instertion (with optional SDI
STRESS Toolkit) details• Reporting of SDI-STRESS pathologicalinsertion statistics• Moving test patterns (bouncing box)
• Import/display TIFF images

SDI† Audio Generation• Choice of fixed tones or chromatic scale –SDI† Pathological
Generation• Choice of fixed or ramp levels – to assist• Conventional SDI pathological stressto assist channel identification

channel identification• Custom config of number of active audiogroups and channels• Master gain control• ST 2022-6: 32 channel audio generation
can be replicated in all sub frames providing
a total of up to 128 channels

patterns, Eq, PLL and CheckField• New proposed SMPTE combined

pathological stress pattern: Eq + PLL + Color
Bars + Clock• Define a percentage combination of SMPTE
or SDI pathological and conventional
patterns up to full frame

2110 Video/ANC
Generation

2110 Audio Generation

SDI† Configuration Dialog

as a flow• 2110: Monitor (GUI) as a flow
• 2110-20: 2K/HD, 4K/UHD video flow
generator (422/444, YCbCr/RBG, 10/12-bit)• 2110-40: 1 x ANC flow generator• Timecode Generator ATC_LTC, ATC_VITC,
locked to PTP or Local Time with Jam Sync
and Drop Frame, VITC1/2 Reverse andsignaling of SDI Line number and H Offset• Import of TIFF images• Bouncing Box pattern movement• ST 2110-20 EUHD 47.95-60p RGB YCbCr
444 formats[PHQXLO-EUHD / PHQXPO-EUHD]• 2110: Generate up to four ST 2110/2022-7
audio flows• 2110-30/-31: Up to:• 12G/6G/3G/1.5G 4K/UHD and 2K/HD SDI• 2110: Generate ST 2110/2022-7 Test Signals

80 audio channels 2110-30 at 125 μs
60 audio channels 2110-31 at 125 μs
10 audio channels 2110-30 at 1 ms
7 audio channels 2110-31 at 1 ms

signal generation• Support for Single, Dual, Quad link SDI

formats. Square division, 2SI, Level A & B• 422, 444, 4224 and 4444, YCbCr and RGB
formats, 10/12 bit• Supports the detection and analysis of
SMPTE Full Range video standards† Requires PHQXL01-3G / PHQXL01E-3G or PHQXP01-3G / PHQXP01E-3G

www.leaderphabrix.com

15

Optional Toolsets

GEN

IP

Audio and Video Generation [PHQXLO-GEN / PHQXPO-GEN]
IP Transmit - Tx Status
Current Total Bandwidth
Used on SFP E Interface

Current Total Bandwidth
Used on SFP F Interface

Active PTP
Reference• At-a-glance status overview of all flows
being encapsulated and transmitted

Transmit Flow
Status Tabs

• Simultaneously transmit two different typesof flow from the unit: Generator Flows andMonitor (GUI) Flows• Displays a summary of the current status of
all selected generator / monitor video, audio
and ancillary flows being transmitted• Use this tab as an overview of all flows
actively being transmitted from the unit,
together with the active PTP reference andan indication of bandwidth used by eachstream and the total bandwidth used oneach interface• Displays the current information about thetest pattern VID, AUD, ANC and monitor VID
and AUD flows

Summary of
SFP E Tx Flows

Summary of
SFP F Tx Flows

Individual Flow
Bandwidth

Tx packet counters

IP Transmit - VID, AUD1-2, AUD3-4, ANC, MON Status• The VID tab displays the active settings for

the Video Generator: Protocol, Interface,Video Source, Packet Read Schedule,
Packing Mode, TR Offset• The AUD1-2, AUD3-4 tabs shows the active
settings for the transmitted audio flows:
Protocol, Packet Time, Channels, Audio
Source• The ANC tab displays the active settings
for the Video Generator flows: Protocol,
Interface, Packet Packing, Keep Alive,
Timecode, TR Offset• The MON tab displays the active settings
for transmission of the Monitor flows:
Protocol, Interface, Video Source, Packet
Read Schedule, Audio Source, Packet Time,
Channels,

Displays the Active
Settings for the Selected
Flows

Transmission Configuration
Configure Generator/
Monitor Flows

Configuration
Options
Summaryof Current
Configuration

16

QxL / QxP

Flow Enable/Disable• List of available flows in an expandable list
• Each minimized flow provides a single

line summary of the current settings for
information• Configure the VID, AUD1, AUD2, AUD3,
AUD4 Generator Flows• Configure the VID MON, AUD MON Monitor
flows• 2110-20: Gapped/Linear Packet Read
Schedule, BPM/GPM Packing Mode
• SDI/Egress Time Stamp, user control of TR
Offset• 2110-40 ANC, Keep Alive and ATC-LTC or
ATC-VITC Timecode locked to PTP or Local
TimeOptional Toolsets
Dolby® E Decoder and Metadata Analyzer [PHQXL0-DOLBY / PHQXP0-DOLBY]The Dolby E Decoder and Metadata Analyzer option provides a clear and accessible view of the Dolby E metadata present
in a selected Dolby E or ED2 audio stream. It also enables you to check the correct timing of Dolby E packets in the audio
signal in an SDI or ST 2022-6 broadcast chain. You can check whether the Dolby E is created correctly and transferredtransparently by network equipment such as routers, switchers, satellite links, etc. You can also choose to monitor the Dolby®
audio from any of the SDI/2022-6 embedded audio, 2110-30/-31 or AES inputs. The decoded output and downmix can bemetered, monitored, Loudness measured, and routed to AES outputs.

Dolby Metadata Analyzer
Source Pair/Channel ofthe Dolby Bitstream
Dolby Begin/End Gain
Values for Each Source
SignalProgram-dependent
Metadata for Dolby E
Program 1
Program-dependent
Metadata for Dolby E
Program 2• Displays the Dolby E metadata present in

the selected Dolby E or ED2 audio stream• Enables you to check the correct timingof embedded Dolby E and ED2 in SDI and
2022-6 payloads• Check that the Dolby E metadata has beencreated correctly for multiple programs
using the easy to read metadata display• You can choose to monitor the Dolby audio
from any of the SDI, ST 2022-6 or 2110
input embedded audio pairs/channels or
the AES input• Dolby stream CRC error detection anddisplay

Dolby Detection in Audio MeteringGroup Pair Labels
Stereo Pair Selectors
Mono Channel Solo
Bus Selectors

Detect any Numberof Dolby E, ED2, DD
and DD+ Pairs
Numeric Values ofEach Meter (dBFS)

Dolby Decoder Metering and Status
8 Channels
AutomaticallyIdentified from Dolby
Program Metadata
2 Channels for 2.0
Stereo Downmix
of Selected Dolby
Program
8 Channel Audio
Metering for Dolby E,
and 2 Channels for
Downmix• Displays 16 audio meters together withpeak level indicators and indication ofaudio pair correlation• Dolby E, Dolby D and Dolby D+ streams
are detected by the system with Dolby
stream presence indicated in blue
• For an SDI or ST 2022-6 signal carrying
embedded Dolby E audio, the Dolby E
timing line number is also displayed below
the detector, either as an absolute value or
relative to the Ideal line number specified
for that video standard

Correlation Indicator for
Each Audio Pair
Dolby Timing Line
Number Absolute or
Relative to Ideal• When the Dolby E decoder is selected as

the metering source (ST 2110, SDI or 20226 input), the view of the analyzer changes
to display the 8 channels of decoded
Dolby E audio as well as the stereo 2.0
downmix• The meter channel identification is
automatically configured from the Dolby
program metadata• Display of Dolby E source, line positioning(SDI, 2022-6), dialogue level and downmix
program source

Dolby Decoder Paneland Status Informationwww.leaderphabrix.com

17

IP PCAP

Optional Toolsets
10G/25G PCAP Tool [PHQXLO-IP-PCAP / PHQXPO-IP-PCAP]

This Packet Capture (PCAP) tool provides a flexible range of options for your capture of the live IP traffic on either a single orboth Media interfaces per IP input (ST 2110). The PCAP data is then saved to USB memory stick for offline analysis using thirdparty network analysis tools. The PCAP data on the USB stick can be accessed remotely via Web Browser.

• Full line-rate capture at 25 Gbps on a single interface, back-to-back packets• Capture data on one or both media interfaces simultaneously up to 50 Gbps
• User control of packet capture size - e.g. Full payload or headers only with user control of the
Packet Capture size (12-1518 Octets)• Manual Start-Stop, Auto Start-Stop at specified time, Capture Start Delay• User controls for auto stop: No of Packets, File size, Duration• Saves to USB stick with the option of Browser File transfer off the unit with remote access
control of USB ports• 4 GB PCAP max. file size

IP Network Traffic Measurement [PHQXLO-IP-MEAS / PHQXPO-IP-MEAS]
An advanced engineering suite of tools for ST 2110 analysis and debug offers theprovision of up to four simultaneous dual Packet Interval Timing measurement
windows per input for easy visualization of network congestion and sender packet
distribution with max, mean and min inter-packet arrival times. Also included isdetailed data reporting of flow packet, clock rates and PTP timing relationship, as
well as the measurements of the ST 2110-21 Network Compatibility model (CINST) and
Virtual Receiver Buffer Model (VRX). Advanced measurement of IP flow latency and
RTP clock timing relationships for debug of Audio, Video and ANC alignment, source
PTP and encapsulation are featured.

Inter-packet Timing• Stream health reporting using histogram to

Advanced Media Timing - Video• TIme of First Received Packet of a Frame (video timing)• Receiver Buffer Margin with respect to TROdefault• Sender RTP offset

show the distribution of inter-packet arrival
times• Simultaneous reporting of ST 2022-7
primary and secondary flow
• Packet counts (log or linear scales) mapped
against arrival times (μs)
• Easy diagnosis of congestion with max,mean and min inter-packet arrival times

ST 2110-21

Advanced Media Timing - Media Latency• Numerical display of Mean, Min and Max latency measurements• Measured RTP clocks/s, Timestamps/s and RTP clocks/timestamp interval• Numerical display of Video to Audio and ANC relative latency measurement

18

QxL / QxP• ST 2110-21 measurement of Network

Compatibility Model (CINST) and Virtual
Receiver Buffer Model (VRX)• User control of VRX buffer read-scheduletiming• User control of CINST buffer drain rate

IP

Optional Toolsets
Packet Interval Profile Generator[PHQXLO-IP-NGT / PHQXPO-IP-NGT](Also Requires PHQXLO-GEN or PHQXPO-GEN)

4K/UHD ST 2110 Extended UHD
Format Support[PHQXLO-EUHD / PHQXPO-EUHD]

A ST 2022-6 packet generation tool for evaluating the abilityof a receiver to handle a ST 2022-6 flow with jitter. Simulate
IP video network packet jitter under a variety of network
conditions by adjusting the transmission distribution profile.
View the interval timing distribution of generated packets,and the number of packets generated each second, against
the deviationof each packet
interval from theexpected interval
time.

(Also Requires PHQXLO-UHD or PHQXPO-UHD)

IP Transmit (ST 2022-6)

EUHD Format Support• Configuration of Transmission flow addresses, port numbers and
SSRC• Inter-packet jitter onto outgoing flow
• Gaussian or uniform distribution• Flow control on/off

HDR

Out of the box the QxL and QxP support YCbCr 4:2:2 andYCbCr/RGB 444 formats in 2110-20 up to a max payload of
approx 12 Gbps. If you are working with Extended Reality (xR),
fixed installation LED walls and Graphics Card applications,then the PHQXLO-EUHD / PHQXPO-EUHD options provide
support for Analysis
and Generation of
UHD/4K YCbCr/RGB
444 formats in therange 47.95P – 60P.

• Analysis of 2110-20 flows at UHD/4K 444 (RGB/YCbCr) 8/10/12 bit
47.95P-60P• 4K60P RGB:12 Mean bandwidth approx. 20 Gbps (equivalent to apeak bandwidth of around 21 Gbps for a gapped flow)

High Dynamic Range (HDR) Visualization & Analysis Toolset[PHQXLO-HDR or PHQXPO-HDR]

The Qx Series’ comprehensive HDR toolset includes a signal generator, CIE chart, Luma false color highlighting or heat map,
waveform monitor and vectorscope. All the main live production SDR and HDR formats are supported: Standard Dynamic
Range (SDR) BT.709, BT.2020 as well as HDR BT.2100 HLG, PQ and Sony S-Log3 and SR Live. The Waveform provides a
Cd/m2 (nits) graticule along with BT.2048 diffuse white markers. The flexible user controlled HDR heatmap offers 7
simultaneous programmable color overlay bands with presets for HDR and SDR ranges, plus a user custom preset. The CIE
1931 xy display provides overlays for BT.709, BT.2020 and ST.2086 gamut (P3) to enhance the visualization and analysis ofyour HDR / WCG content.
HDR Generator
An extensive set of test patterns include BT.2111 HDR color bars for HLG, PQ and SR Live as well as a full set of SDR 709
patterns mapped via display light to each of the four HDR formats for line checks, comparative monitor set-up and theevaluation of HDR to SDR converters.

False Color Highlighting• Programmable Heat Map to highlightluminance zones providing quick
identification of shadows, skin or mid-tones
or specular highlights• Seven simultaneous programmable color
overlay bands• Presets for HDR and SDR ranges plus user
customAnalyzer - CIE Chart• CIE 1931 xy display• Single line mode linked to picture cursor• Pan and zoom
• ITU-R BT. 709, BT. 2020 and ST 2086 gamut

HDR Waveform• Waveform HDR graticules with Nits (Cd/m2)• BT. 2408 diffuse white markers• Support for BT. 1886, BT. 2100 HLG and PQ,
Sony S-Log3, SR Liveoverlays• Tooltip co-ordinate display• Support for BT. 1886, BT. 2100 HLG and PQ,
Sony S-Log3, SR Livewww.leaderphabrix.com

19

QxP
Qx

Specifications

QxL

Formats supported (Generation, Analysis & Monitoring)

ST 2110-20/-30/-31/-40 / 2022-7 / 2022-6 Analysis over 10G Ethernet
ST 2110-20/-30/-31/-40 / 2022-7 / 2022-6 Analysis over 25G Ethernet-

ST 2110-20/-30/-31/-40 / 2022-7 Generation-

ST 2022-6 Generation
SDI IO

Factory Option

Factory Option

3G / 1.5G / 270M*-SDI HD / SD Analysis

Factory Option

Factory Option

3G / 1.5G / 270M*-SDI HD Generation
12G / 6G / 3G / 1.5G-SDI UHD Over SDI
25G IP Link Rates Over SFP28

Hardware and Software Options Supported
Audio / Video Generator (SDI, ST 2022-6, ST 2110)

-

†

†

†

†

(SDI, 2022-6)
Factory Option(SDI, 2022-6, 2110)
Factory Option(SDI, 2022-6, 2110)
Factory Option(SDI)(SDI, 2110)(SDI, 2110)

4 x SDI inputs, SD / HD / 3G, 75 ohm terminated BNC

Factory Option

Factory Option

2 x SFP+ MSA / Non-MSA cages (12 Gbps copper or fiber SDI interfaces)

Factory Option

Factory Option

4 x SDI outputs, SD* / HD / 3G, 75 ohm BNC

Factory Option

Factory Option--

RTE™ Real-Time Eye input (12G/6G/3G/HD/SD-SDI) x 1 (SDI input A) BNC
UHD / 4K Upgrade
SDI-STRESS Testing Toolset (Requires SDI Eye and Jitter Toolset)
Data View Analyzer, ANC Status and ANC Inspector
32 Channel Audio Metering and 5.1/2.0 Loudness Measurement
HDR/WCG Support
Dolby E Analysis
ST 2022-6, ST 2110/20/30/31/40 Decap with Class C Audio, ST 2022-7, Dual PTP
ST 2110 Network Traffic Measurement Toolset
ST 2022-6 Network Traffic Generator Toolset
ST 2110-20/30/31/40 Generator Toolset with Class C Audio, ST 2022-7-

PCAP
EUHD Formats over 25G 2110-20-

SDI inputs / outputs

Ethernet inputs / outputs (accepts generic SFPs)
2 x SFP+ 10G Cages (shared with SDI SFP+ MSA/Non-MSA cages)
2 x SFP28 10 / 25G cages-

2 x QSFP28 cages (Reserved)-

Audio inputs / outputs

4 x 75 ohm AES selectable I/O (26 pin high-density D-Type socket)
1 x Stereo analog audio output (26 pin high-density D-Type socket)
8 channel 48 kHz PCM audio on HDMI and SDI Instrument output

User interface

Integrated 1920 x 1200 7 inch LCD multitouch touchscreen--

HDMI instrument output, 1920 x 1080, 4:4:4 RGB, Type A

HDMI 1.4

HDMI 2.0a

HDMI 2.0a

BNC

Micro BNC

Micro BNC--SDI 3Gbit instrument out, 1920 x 1080, 4:2:2 YCbCr
ST 2110-20, ST 2110-30 instrument out, 1920 x 1080, 4:2:2 YCbCr-

Remote Browser GUI access (noVNC)

Reference

2 x 75 ohm BNC looping reference input, tri-level or B&B with cross lock1 x 75 ohm Micro-BNC reference input, tri-level or B&B with cross lockNetworking & control-

10/100/1000 BASE-T

8 x bi-directional GPI (26 pin high-density D-Type socket)

Monitoring

Integral Speaker / Headphone Socket (1/4”)--

211 x 44 x 253 mm

211 x 44 x 253 mm

215 x 132 x 330 mm

1.9 kg

1.9 kg

4.1 kg ‡10-18V, 50W typical, 100W max

10-18V, 70W typical, 100W max

11-17V, 70W typical, 100W max

90-264 VAC, 50 W typical ,120 W max

90-264 VAC, 70 W typical,
120 W max-

Internal AC power supply with IEC connector--

100-240 VAC, 77 typical,
110 W max

Choice of External Battery V-mount or Gold-mount--Form factor

Size (Width x Height x Depth - excluding projections)
WeightElectrical
4 pin XLR DC power connector
External power supply

Warranty

Warranty (1 year)
Extended Warranty Package (3 - 5 years)

20

QxL / QxP

† Requires SDI Factory Option‡ Excludes external battery

Standard

Optional

Rear Panel - IO View
Power
10 - 18 V QxL
11 - 17 V QxP

IP Only

Networking(10/100/1000
Base-T)

Ref
User HDMI2.0b
I/O
Interface Instrument
USB Output (to1080p60) SDI
2.0
InstrumentOut(to 1080p60)

2 x SFP28(25GE, 10GE)

2 x QSFP28(100GE, 50GE, 40GE)(For future use)

Stereo Audio Out,
8x GPI I/O,
4x AES I/OWith Factory-fitted SDI Option
Generator, Generator Copy or
SDI Input Loop Out
12G/6G/3G/HD/SD-SDI BNC x4

Networking(10/100/1000
Base-T)

User
Interface
USB
2.0

Ref
I/O
SDI
Instrument Out(to 1080p60)

12G/6G/3G/HD/SD*-SDI
Eye
Analyzer SDI Inputs(SFP+) MSA/NON-MSA
12G/6G/3G/HD/SD*-SDI
12 Gbps Copper or Fiber
BNC x42 x SFP28(25GE, 10GE)

2 x QSFP28(100GE, 50GE, 40GE)(For future use)Power
10 - 18 V QxL
11 - 17 V QxPStereo Audio Out,
8x GPI I/O,
4x AES I/O

HDMI2.0b
Instrument
Output(to 1080p60)

www.leaderphabrix.com

21Ordering QxL
QxL Chassis Options

QxL SDI/IP Software Options

PHQXL

QxL 1U SD/HD/2K 10GbE IP Rasterizer, Analyser
only

PHQXLO-DOLBY

Dolby E Decoder, Metadata Analyser, LtRt/LoRo
downmix, metering

PHQXL01-3G

QxL 1U SD/HD/2K 10GbE IP/SDI Rasterizer,
Analyser only

PHQXLO-GEN

SDI/IP AV Test Signal Generator(SDI requires PHQXL01-3G or PHQXL01E-3G)

PHQXL01E-3G

QxL 1U SD/HD/2K 10GbE IP/SDI Rasterizer, Eye/
Jitter, Analyser only

PHQXLO-UHD

2K Extended + UHD/4K IP/SDI(ST 2110 requires PHQXLO-IP-25G)(SDI requires PHQXL01-3G or PHQXL01E-3G)

PHQXLO-HDR

HDR/WCG, CIE1931, HDR Heat map (PQ, HLG,
S-Log3, SR Live)QxL Chassis Upgrade Options
PHQXLM-01

QxL SDI I/O return to factory upgrade(requires PHQXL)PHQXLM-01E

QxL SDI Eye/Jitter return to factory upgrade(requires PHQXL01-3G)

QxL IP Options
PHQXLO-IP-25G

25GbE media network(requires 2x PHSFP-25G-SR or PHSFP-25G-LR)

PHQXLO-EUHD

Adds RGB, 12b, 444, 48-60Hz formats to ST2110(requires PHQXLO-UHD)

PHQXLO-IP-MEAS

IP Measurement 2110-21, PIT histograms, timing

PHQXLO-IP-PCAP

PCAP 2x25Gbps line rate capture tool, 4GB max.

PHQXLO-IP-NGT

2022-6 IP Network traffic Generator Tool
(requires PHQXLO-GEN)

PHSFP-10GE-SR

SFP+ 10GBASE-SR Ethernet MM 850nm 300m

PHSFP-10GE-LR

SFP+ 10GBASE-LR Ethernet SM 1310nm 10km

PHSFP-25GE-SR

SFP28 25GBASE-SR Ethernet MM 850nm 100m

PHSFP-25GE-LR

SFP28 25GBASE-LR Ethernet SM 1310nm 10km

QxL SDI Options
PHQXLO-SDISTRESS

12G-SDI Stress Test Toolset (requires PHQXL01E3G, PHQXLO-UHD, PHQXLO-GEN)

PHSFP-RT12-1310

SFP+ SM(10km) LC Non-MSA, Tx 1310nm, Rx 12601620nm 12G/6G/3G/HD/SD*-SDI

QxL Fitting Kits / Cables
PHQXC-1

12G-SDI Eye Measurement Test Cable 1m

PHQXK119 inch rackmount kit (1x Qx/QxL chassis)

PHQXK219 inch rackmount kit (2x Qx/QxL chassis)

PHQXK39.5 inch rackmount kit (1x Qx/QxL chassis)

QxL Extended Warranty
PHQXL-3YEARPHQXL Upgrade from 1 to 3 Year Warranty(excludes SFP)

PHQXL-5YEARPHQXL Upgrade from 1 to 5 Year Warranty(excludes SFP)

PHQXL01-3YEARPHQXL01 Upgrade from 1 to 3 Year Warranty(excludes SFP)

PHQXL01-5YEARPHQXL01 Upgrade from 1 to 5 Year Warranty(excludes SFP)

PHQXL01E-3YEARPHQXL01E Upgrade from 1 to 3 Year Warranty(excludes SFP)

PHQXL01E-5YEARPHQXL01E Upgrade from 1 to 5 Year Warranty(excludes SFP)

* Upcoming software release22

QxL / QxPOrdering QxPQxP Chassis Options

QxP SDI/IP Software Options

PHQXP-VQxP 3U SD/HD/2K 10GbE IP Waveform Monitor/
Analyser, V-mount

PHQXPO-DOLBY

Dolby E Decoder, Metadata Analyser, LtRt/LoRo
downmix, metering

PHQXP-GQxP 3U SD/HD/2K 10GbE IP Waveform Monitor/
Analyser, G-mount

PHQXPO-GEN

SDI/IP AV Test Signal Generator(SDI requires PHQXP01-3G or PHQXP01E-3G)

PHQXP01-3G-V

QxP 3U SD/HD/2K 10GbE IP/SDI Waveform Monitor/Analyser, V-mount

PHQXPO-UHD

2K Extended + UHD/4K IP/SDI(SDI requires PHQXP01-3G or PHQXP01E-3G)

PHQXP01-3G-G

QxP 3U SD/HD/2K 10GbE IP/SDI Waveform Monitor/Analyser, G-mount

PHQXPO-HDR

HDR/WCG, CIE1931, HDR Heat map (PQ, HLG,
S-Log3, SR Live)

PHQXP01E-3G-V

QxP 3U SD/HD/2K 10GbE IP/SDI Waveform Monitor/Analyser, Eye/Jitter, V-mount

PHQXP01E-3G-G

QxP 3U SD/HD/2K 10GbE IP/SDI Waveform Monitor/ Analyzer, Eye/Jitter, G-mountQxP SDI Options
PHQXPO-SDISTRESS

12G-SDI Stress Test Toolset (requires PHQXP01E3G, PHQXPO-UHD, PHQXPO-GEN)QxP Chassis Upgrades (Return to Factory)

PHSFP-RT12-1310

SFP+ SM(10km) LC Non-MSA, Tx 1310nm, Rx 12601620nm SD*/HD/3G/6G/12G-SDI

PHQXPM-01

QxP SDI I/O return to factory upgrade(requires PHQXP)QxP Fitting Kits

PHQXPM-01E

QxP SDI Eye/Jitter return to factory upgrade(requires PHQXP01-3G)

QxP IP Options
PHQXPO-IP-25G

25GbE media network(requires 2x PHSFP-25G-SR or PHSFP-25G-LR)

PHQXPO-EUHD

Add RGB, 12b, 444, 48-60Hz formats to ST2110(requires PHQXPO-UHD)

PHQXPO-IP-MEAS

IP Measurement 2110-21, PIT histograms, timing

PHQXPO-IP-PCAP

PHQXC-1

12G-SDI Eye Measurement Test Cable 1m

PHQXK7

3U 19 inch rackmount kit (1x QxP Chassis)

PHQXK8

3U 19inch rackmount kit (2x QxP Chassis)

PHQXK9QxP desktop kit (adjustable feet plus handle)

QxP Extended Warranty
PHQXP-3YEARPHQXP Upgrade from 1 to 3 Year Warranty(excludes SFP)

PCAP 2x25Gbps line rate capture tool, 4GB max.

PHQXP-5YEARPHQXPO-IP-NGT

2022-6 IP Network traffic Generator Tool
(requires PHQXPO-GEN)

PHQXP Upgrade from 1 to 5 Year Warranty(excludes SFP)

PHQXP01-3YEARPHSFP-10GE-SR

SFP+ 10GBASE-SR Ethernet MM 850nm 300m

PHQXP01 Upgrade from 1 to 3 Year Warranty(excludes SFP)

PHSFP-10GE-LR

SFP+ 10GBASE-LR Ethernet SM 1310nm 10km

PHQXP01-5YEARPHQXP01 Upgrade from 1 to 5 Year Warranty(excludes SFP)

PHSFP-25GE-SR

SFP28 25GBASE-SR Ethernet MM 850nm 100m

PHQXP01E-3YEARPHQXP01E Upgrade from 1 to 3 Year Warranty(excludes SFP)

PHSFP-25GE-LR

SFP28 25GBASE-LR Ethernet SM 1310nm 10km

PHQXP01E-5YEARPHQXP01E Upgrade from 1 to 5 Year Warranty(excludes SFP)

* Upcoming software releasewww.leaderphabrix.com

23

SDI SFP Interfaces[Requires PHQXL01-3G / PHQXP01-3G or PHQXL01E-3G / PHQXP01E-3G]
SDI SFP Interface

Link TypeSFP+B Link Rates

SFP+A Link Rates

SDI Transceivers Only
SFP Interface

N/A

Rx Ch1

Tx Ch1
BNC A Tx
0.27*/1.5/3/6/12

SIngle Link: Rx/Tx

N/A

N/A

BNC A Rx
0.27/1.5/3/6/12

Dual Link: N/A

N/A

N/A

N/A

N/A

Quad Link: N/A

N/A

N/A

N/A

N/A

SFP Interface

Rx Ch1

Tx Ch1Rx Ch1

Tx Ch1

Single Link: Rx/Tx

N/A

BNC C Tx
(Tx Copy) 0.27*/1.5/3/6/12

BNC A Rx
0.27/1.5/3/6/12

BNC A Tx
0.27*/1.5/3/6/12

Dual Link: Rx/Tx

BNC C Rx
0.27*/1.5/3/6

BNC C Tx
0.27*/1.5/3/6

BNC A Rx
0.27/1.5/3/6

BNC A Tx
0.27*/1.5/3/6

Quad Link: N/A

N/A

N/A

N/A

N/A

SFP Interface

N/A

N/A

Rx Ch1Rx Ch2Single Link: Rx

N/A

N/A

BNC A Rx
0.27/1.5/3/6/12

N/A
BNC B Rx
0.27*/1.5/3/6

One SDI Transceiver in Cage A

Two SDI Transceivers in Cages
A&B

N/A

SDI Dual Receivers Only

One SDI Dual Receiver in Cage A

Two SDI Dual Receivers in Cages
A&B

Dual Link: Rx

N/A

N/A

BNC A Rx
0.27/1.5/3/6

Quad Link: N/A

N/A

N/A

N/A

N/A

SFP Interface

Rx Ch1Rx Ch2Rx Ch1Rx Ch2Single Link: Rx

N/A

N/A

BNC A Rx
0.27/1.5/3/6/12

N/A
BNC B Rx
0.27*/1.5/3/6
BNC B Rx
0.27*/1.5/3

Dual Link: Rx

N/A

N/A

BNC A Rx
0.27/1.5/3/6

Quad Link: Rx 1 2

BNC C Rx
0.27*/1.5/3

BNC D Rx
0.27*/1.5/3

BNC A Rx
0.27/1.5/3

SFP Interface

N/A

N/A

Tx Ch2Tx Ch1
BNC A Tx
0.27*/1.5/3/6/12

SDI Dual Transmitters Only

One SDI Dual Transmitter in Cage A

Single Link: Tx

N/A

N/A

BNC B Tx
(Tx Copy) 0.27*/1.5/3/6

Dual Link: Tx

N/A

N/A

BNC B Tx
0.27*/1.5/3/6

BNC A Tx
0.27*/1.5/3/6

Quad Link: N/A

N/A

N/A

N/A

N/A

SFP Interface

Tx Ch2Tx Ch1

Tx Ch2Tx Ch1

Single Link: Tx

BNC D Tx
(Tx Copy) 0.27*/1.5/3/6

BNC C Tx
(Tx Copy) 0.27*/1.5/3/6/12

BNC B Tx
(Tx Copy) 0.27*/1.5/3/6

BNC A Tx
0.27*/1.5/3/6/12

Dual Link: Tx

BNC D Tx
(Tx Copy) 0.27*/1.5/3/6

BNC C Tx
(Tx Copy) 0.27*/1.5/3/6

BNC B Tx
0.27*/1.5/3/6

BNC A Tx
0.27*/1.5/3/6

Quad Link: Tx 2 3

BNC D Tx
0.27*/1.5/3

BNC C Tx
0.27*/1.5/3

BNC B Tx
0.27*/1.5/3

BNC A Tx
0.27*/1.5/3

SFP Interface

Rx Ch1Rx Ch2Tx Ch2Tx Ch1

Single Link: Rx/Tx

BNC C Rx
0.27*/1.5/3/6/12 4

N/A

BNC B Tx
(Tx Copy) 0.27*/1.5/3/6

BNC A Tx
0.27*/1.5/3/6/12

Dual Link: Rx/Tx

BNC C Rx
0.27*/1.5/3/6

BNC D Rx
0.27*/1.5/3/6

BNC B Tx
0.27*/1.5/3/6

BNC A Tx
0.27*/1.5/3/6

Quad Link: N/A

N/A

N/A

N/A

N/A

Two SDI Dual Transmitters in Cages
A&B

SDI Dual Transmitter plus SDI Dual Receiver

One SDI Dual Transmitter(Cage A) and One Dual SDI
Receiver (Cage B)

1 In quad link 2SI the Receivers will auto adapt to any order of sub-image to BNC mapping.
2 In quad link square division the sub image order is: BNC A:TL, BNC B:TR, BNC C:BL, BNC D:BR.
3 In quad link 2SI the sub image order is: BNC A:Sub 1, BNC B:Sub 2, BNC C:Sub 3, BNC D:Sub 4.
4 SD-SDI (270M (0.27G)) video is not supported for Mixed Dual Transmitters and Receivers

24

QxL / QxP

* Upcoming Software ReleaseSupported 2K/HD/SD SDI Formats
The following SDI formats are available on QxL and QxP.
SMPTE Stnds.
Link (Content)

Interface

Resolution

Sampling
Structure

Pixel
Depth

ST 259 (ST 125)

SD (625i)

720 x 576

4:2:2 (YCbCr)

10

Frame/Field Rate

HDR†

SDI‡

2022-6

50i

-

OA

A

-

OA

A

59.94i

ST 259 (ST 125)

SD (525i)

720 x 485

4:2:2 (YCbCr)

10

ST 292 (ST 296)

HD

1280 x 720

4:2:2 (YCbCr)

1060p, 59.94p, 50p, 30p, 29.97p, 25p,

O●

O●

●

O●

O●

●

ST 292 (ST 274)

HD

1920 x 1080

4:2:2 (YCbCr)

1060i, 59.94i, 50i
30p, 29.97p, 25p, 24p, 23.98p

ST 292 (RP 211)

HD

1920 x 1080

4:2:2 (YCbCr)

10

30psF, 29.97PsF, 25psF, 24PsF, 23.98PsF

O●

O●

●

4:2:2 (YCbCr)

1030p, 29.97p, 25p, 24p, 23.98p,
30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF

O●

O●

●

ST 292 (ST 2048-2)
ST 372 (ST 274)

HD

2048 x 1080

Dual Link HD

1920 x 1080

4:2:2 (YCbCr)

1060p, 59.94p, 50p

O●

O●

-

10

60i, 59.94i, 50i, 30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

O●

-

12

60i, 59.94i, 50i
30p, 29.97p, 25p, 24p, 23.98p

O●

O●

-

4:2:2 (YCbCr)

1260i, 59.94i, 50i, 30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

O●

-

10

60p, 59.94p, 50p, 48p, 47.95p

O●

O●

-

10

30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

O●

-

ST 372 (ST 274)

Dual Link HD

1920 x 1080

4:4:4 (YCbCr/RGB)
4:4:4:4 (YCbCrA/RGBA)

ST 372 (ST 274)

Dual Link HD

1920 x 1080

4:4:4 (YCbCr/RGB)

ST 372 (ST 274)

Dual Link HD

1920 x 1080

ST 372 (ST 2048-2)

Dual Link HD

2048 x 1080

4:2:2 (YCbCr)

2048 x 1080

4:4:4 (YCbCr/RGB)
4:4:4:4 (YCbCrA/RGBA)

ST 372 (ST 2048-2)
ST 372 (ST 2048-2)

Dual Link HD
Dual Link HD

2048 x 1080

4:4:4 (YCbCr/RGB)

12

30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

O●

-

12

30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

O●

-

ST 372 (ST 2048-2)

Dual Link HD

2048 x 1080

4:2:2 (YCbCr)
4:2:2:4 (YCbCrA)

ST 425-1 (ST 274)

3G Level A (1)1920 x 1080

4:2:2 (YCbCr)

1060p, 59.94p, 50p

O●

O●

●

ST 425-1 (ST 2048-2)

3G Level A (1)2048 x 1080

4:2:2 (YCbCr)

1060p, 59.94p, 50p, 48p, 47.95p

O●

O●

●

ST 425-1 (ST 296)

3G Level A (2)1280 x 720

4:4:4 (YCbCr/RGB)
4:4:4:4 (YCbCrA/RGBA)

1060p, 59.94p, 50p, 30p, 29.97p

O●

O●

●

1920 x 1080

4:4:4 (YCbCr/RGB)
4:4:4:4 (YCbCrA/RGBA)

1060i, 59.94i, 50i, 30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

O●

●

10

30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

O●

●

12

60i, 59.94i, 50i, 30p, 29.97p, 25p, 24p, 23.98p

O●

O●

●

12

30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

O●

●

ST 425-1 (ST 274)

3G Level A (2)ST 425-1 (ST 2048-2)

3G Level A (2)2048 x 1080

4:4:4 (YCbCr/RGB)
4:4:4:4 (YCbCrA/RGBA)

ST 425-1 (ST 274)

3G Level A (3)1920 x 1080

4:4:4 (YCbCr/RGB)

ST 425-1 (ST 2048-2)
ST 425-1 (ST 274)

3G Level A (3)3G Level A (4)2048 x 1080

4:4:4 (YCbCr/RGB)

1920 x 1080

4:2:2 (YCbCr)

1260i, 59.94i, 50i, 30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

O●

●

12

30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

O●

●

60p, 59.94p, 50p

O●

O●

●

ST 425-1 (ST 2048-2)

3G Level A (4)2048 x 1080

4:2:2 (YCbCr)
4:2:2:4 (YCbCrA)

ST 425-1 (ST 274)

3G Level B-DL (I)1920 x 1080

4:2:2 (YCbCr)

10

ST 425-1 (ST 2048-2)

3G Level B-DL (I)2048 x 1080

4:2:2 (YCbCr)

1060p, 59.94p, 50p, 48p, 47.95p

O●

O●

●

1920 x 1080

4:4:4 (YCbCr/RGB)
4:4:4:4 (YCbCrA/RGBA)

1060i, 59.94i, 50i, 30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

O●

●

10

30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

O●

●

ST 425-1 (ST 274)

3G Level B-DL (II)ST 425-1 (ST 2048-2)

3G Level B-DL (II)2048 x 1080

4:4:4 (YCbCr/RGB)
4:4:4:4 (YCbCrA/RGBA)

ST 425-1 (ST 274)

3G Level B-DL (III)1920 x 1080

4:4:4 (YCbCr/RBG)

1260i, 59.94i, 50i, 30p, 29.97p, 25p, 24p, 23.98p

O●

O●

●

ST 425-1 (ST 2048-2)

3G Level B-DL (III)2048 x 1080

4:4:4 (YCbCr/RBG)

1230PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

O●

●

12

60i, 59.94i, 50i, 30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

O●

●

12

30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF
30p, 29.97p, 25p, 24p, 23.98p

O●

O●

●

ST 425-1 (ST 274)
ST 425-1 (ST 2048-2)

3G Level B-DL (IV)
3G Level B-DL (IV)1920 x 1080

4:2:2 (YCbCr)

2048 x 1080

4:2:2 (YCbCr)
4:2:2:4 (YCbCrA)

KEY● - Generator with PHQXLO-GEN / PHQXPO-GEN option and Analyzer
O - Optional
O● - Optional Generator with PHQXLO-GEN / PHQXPO-GEN option and Analyzer
A - Analyzer Only‘-’ - Not Available† Note: Optional HDR formats require PHQXLO-HDR or PHQXPO-HDR‡ Note: SDI formats require PHQXL01-3G / PHQXP01-3G or PHQXL01E-3G / PHQXP01E-3G

www.leaderphabrix.com

25

Supported 2K/HD/SD IP Formats
The following 2K/HD/SD ST 2110-20 formats are provided as standard.
Resolution

26

Sampling Structure

Pixel
Depth

Frame/Field Rate

2110
HDR†

2110
SDR

720 x 576

4:2:2 (YCbCr)

10

50i

-

A

720 x 485

4:2:2 (YCbCr)

1059.94i-

A

1280 x 720

4:2:2 (YCbCr)

860p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

A1280 x 720

4:2:2 (YCbCr)

1060p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

O●●1280 x 720

4:4:4(YCbCr/RGB)

860p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

A1280 x 720

4:4:4(YCbCr/RGB)

1060p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

O●●

1920 x 1080

4:2:2(YCbCr)

860I, 59.94I, 50I

OA

A

1920 x 1080

4:2:2 (YCbCr)

1060i, 59.94i, 50i

O●●

1920 x 1080

4:2:2(YCbCr)

1260I, 59.94I, 50I

O●●

1920 x 1080

4:4:4(YCbCr/RGB)

8

60I, 59.94I, 50I

OA

A

1920 x 1080

4:4:4(YCbCr/RGB)

10

60I, 59.94I, 50I

O●●

1920 x 1080

4:4:4(YCbCr/RGB)

12

60I, 59.94I, 50I

O●●

1920 x 1080

4:2:2 (YCbCr)

860p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

A

1920 x 1080

4:2:2 (YCbCr)

1060p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

O●●

1920 x 1080

4:2:2 (YCbCr)

1260p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

O●●

1920 x 1080

4:4:4(YCbCr/RGB)

860p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

A

1920 x 1080

4:4:4(YCbCr/RGB)

1060p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

O●●

1920 x 1080

4:4:4(YCbCr/RGB)

1260p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

O●●

1920 x 1080

4:2:2 (YCbCr)

830PsF, 29.97PsF, 25PsF, 24PsF, 23.97PsF

OA

A

1920 x 1080

4:2:2 (YCbCr)

10

30PsF, 29.97PsF, 25PsF, 24PsF, 23.98PsF

O●●

1920 x 1080

4:2:2 (YCbCr)

12

30PsF, 29.97PsF, 25PsF, 24PsF, 23.97PsF

O●●

1920 x 1080

4:4:4(YCbCr/RGB)

8

30PsF, 29.97PsF, 25PsF, 24psF, 23.97PsF

OA

A

1920 x 1080

4:4:4(YCbCr/RGB)

10

30psF, 29.97psF, 25psF, 24PsF, 23.97PsF

O●●

1920 x 1080

4:4:4(YCbCr/RGB)

12

30PsF, 29.97PsF, 25PsF, 24PsF, 23.97PsF

O●●

2048 x 1080

4:2:2 (YCbCr)

860p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

A2048 x 1080

4:2:2 (YCbCr)

1060p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

O●●2048 x 1080

4:2:2 (YCbCr)

1260p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

O●●2048 x 1080

4:4:4(YCbCr/RGB)

860p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

A2048 x 1080

4:4:4(YCbCr/RGB)

1060p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

O●●2048 x 1080

4:4:4(YCbCr/RGB)

1260p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

O●●2048 x 1080

4:2:2(YCbCr)

830PsF, 29.97PsF, 25PsF, 24PsF, 23.97PsF

OA

A

2048 x 1080

4:2:2(YCbCr)

10

30PsF, 29.97PsF, 25PsF, 24PsF, 23.97PsF

O●●

2048 x 1080

4:2:2(YCbCr)

12

30PsF, 29.97PsF, 25PsF, 24PsF, 23.97PsF

O●●

2048 x 1080

4:4:4(YCbCr/RGB)

8

30PsF, 29.97PsF, 25PsF, 24PsF, 23.97PsF

OA

A

2048 x 1080

4:4:4(YCbCr/RGB)

10

30PsF, 29.97PsF, 25PsF, 24PsF, 23.97PsF

O●●

2048 x 1080

4:4:4(YCbCr/RGB)

12

30PsF, 29.97PsF, 25PsF, 24PsF, 23.97PsF

O●

●

KEY
● - Generator with PHQXLO-GEN / PHQXPO-GEN option and Analyzer
O - Optional
O● - Optional Generator with PHQXLO-GEN / PHQXPO-GEN option and Analyzer
A - Analyzer Only‘-’ - Not Available† Note: Optional HDR formats require PHQXLO-HDR or PHQXPO-HDR

QxL / QxP

* Upcoming Software ReleaseSupported 4K/UHD Formats
The following SDI formats are optional on QxL/QxP[PHQXLO-UHD / PHQXPO-UHD + PHQXL01-3G / PHQXP01-3G or PHQXL01E-3G / PHQXP01E-3G]
SMPTE Stnds.
Link (Content)

Interface

Resolution

Sampling
Structure

Pixel
Depth

Frame/Field Rate

SDI
HDR†

SDI
SDR

ST 425-3 Annex B.1 (ST 2036-1)

Quad-link HD-SQ

3840 x 2160

4:2:2 (YCbCr)

1030p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 425-3 Annex B.1 (ST 2048-1)

Quad-link HD-SQ

4096 x 2160

4:2:2 (YCbCr)

1030p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 425-3 Annex B.2, (ST 2036-1)

Dual 3G-B-DS

3840 x 2160

4:2:2 (YCbCr)

1030p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 425-3 Annex B.2, (ST 2048-1)

Dual 3G-B-DS

4096 x 2160

4:2:2 (YCbCr)

1030p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 2081-10 M1 (ST 2036-1)

6G-2SI

3840 x 2160

4:2:2 (YCbCr)

1030p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 2081-10 M1 (ST 2048-1)

6G-2SI

4096 x 2160

4:2:2 (YCbCr)

1030p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 425-5 (ST 2036-1)

Quad-link 3G-A, B (1) 2SI

3840 x 2160

4:2:2 (YCbCr)

1060p, 59.94p, 50p

O●O●

ST 425-5 (ST 2048-1)

Quad-link 3G-A, B (1) 2SI

4096 x 2160

4:2:2 (YCbCr)

1060p, 59.94p, 50p, 48p, 47.95p

O●O●

ST 425-5 (ST 2036-1)

Quad-link 3G-A, B (2) 2SI

3840 x 2160

4:4:4 (YCbCr/RGB)

1030p, 29.97p, 25p, 24p, 23.98p

O●O●4096 x 2160

4:4:4 (YCbCr/RGB)
4:4:4:4 (YCbCrA/RGBA)

1030p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 425-5 (ST 2048-1)

Quad-link 3G-A, B (2) 2SI

ST 425-5 (ST 2036-1)

Quad-link 3G-A, B (3) 2SI

3840 x 2160

4:4:4 (YCbCr/RGB)

1230p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 425-5 (ST 2048-1)

Quad-link 3G-A, B (3) 2SI

4096 x 2160

4:4:4 (YCbCr/RGB)

1230p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 425-5 (ST 2036-1)

Quad-link 3G-A, B (4) 2SI

3840 x 2160

4:2:2 (YCbCr)

1230p, 29.97p, 25p, 24p, 23.98p

O●

O●

12

30p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 425-5 (ST 2048-1)

Quad-link 3G-A, B (4) 2SI

4096 x 2160

4:2:2 (YCbCr)
4:2:2:4 (YCbCrA)

ST 425-5 Annex B (ST 2036-1)

Quad-link 3G-A, B (1) SQ

3840 x 2160

4:2:2 (YCbCr)

1060p, 59.94p, 50p

O●O●

ST 425-5 Annex B (ST 2048-1)

Quad-link 3G-A, B (1) SQ

4096 x 2160

4:2:2 (YCbCr)

1060p, 59.94p, 50p, 48p, 47.95p

O●O●

ST 425-5 Annex B (ST 2036-1)

Quad-link 3G-A, B (2) SQ

3840 x 2160

4:4:4 (YCbCr/RGB)

1030p, 29.97p, 25p, 24p, 23.98p

O●

O●

10

30p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 425-5 Annex B, (ST 2048-1)

Quad-link 3G-A, B (2) SQ

4096 x 2160

4:4:4 (YCbCr/RGB)
4:4:4:4 (YCbCrA/RGBA)

ST 425-5 Annex B (ST 2036-1)

Quad-link 3G-A, B (3) SQ

3840 x 2160

4:4:4 (YCbCr/RGB)

1230p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 425-5 Annex B, (ST 2048-1)

Quad-link 3G-A, B (3) SQ

4096 x 2160

4:4:4 (YCbCr/RGB)

1230p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 425-5 Annex B (ST 2036-1)

Quad-link 3G-A, B (4) SQ

3840 x 2160

4:2:2 (YCbCr)

1230p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 425-5 Annex B (ST 2048-1)

Quad-link 3G-A, B (4) SQ

4096 x 2160

4:2:2 (YCbCr)
4:2:2:4 (YCbCrA)

1230p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 2081-11 M1, ST 425-5 (ST 2036-1)

Dual-link 6G-2SI (I)3840 x 2160

4:2:2 (YCbCr)

1060p, 59.94p, 50p

O●O●

ST 2081-11 M1, ST 425-5 (ST 2048-1)

Dual-link 6G-2SI (I)4096 x 2160

4:2:2 (YCbCr)

1060p, 59.94p, 50p, 48p, 47.95p

O●O●

ST 2081-11 M1, ST 425-5 (ST 2036-1)

Dual-link 6G-2SI (II)3840 x 2160

4:4:4 (YCbCr/RGB)

1030p, 29.97p, 25p, 24p, 23.98p

O●

O●

10

30p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 2081-11 M1, ST 425-5 (ST 2048-1)

Dual-link 6G-2SI (II)4096 x 2160

4:4:4 (YCbCr/RGB)
4:4:4:4 (YCbCrA/RGBA)

ST 2081-11 M1 ST 425-5 (ST 2036-1)

Dual-link 6G-2SI (III)3840 x 2160

4:4:4 (YCbCr/RGB)

1230p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 2081-11 M1, ST 425-5 (ST 2048-1)

Dual-link 6G-2SI (III)4096 x 2160

4:4:4 (YCbCr/RGB)

1230p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 2081-11 M1 ST 425-5 (ST 2036-1)

Dual-link 6G-2SI (IV)3840 x 2160

4:2:2 (YCbCr/RGB)

1230p, 29.97p, 25p, 24p, 23.98p

O●O●4096 x 2160

4:2:2 (YCbCr)
4:2:2:4 (YCbCrA)

1230p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 2081-11 M1 ST 425-5 (ST 2048-1)

Dual-link 6G-2SI (IV)ST 2082-10 M1, ST 425-5 (ST 2036-1)

12G-2SI (I)3840 x 2160

4:2:2 (YCbCr)

1060p, 59.94p, 50p

O●O●

ST 2082-10 M1, ST 425-5 (ST 2048-1)

12G-2SI (I)4096 x 2160

4:2:2 (YCbCr)

1060p, 59.94p, 50p, 48p, 47.95p

O●O●

ST 2082-10 M1 ST 425-5 (ST 2036-1)

12G -2SI (II)3840 x 2160

1030p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 2082-10 M1 ST 425-5 (ST 2048-1)

12G -2SI (II)4096 x 2160

1030p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 2082-10 M1 ST 425-5 (ST 2036-1)

12G-2SI (III)3840 x 2160

4:4:4 (YCbCr/RGB)

1230p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 2082-10 M1 ST 425-5 (ST 2048-1)

12G-2SI (III)4096 x 2160

4:4:4 (YCbCr/RGB)

1230p, 29.97p, 25p, 24p, 23.98p

O●

O●

12

30p, 29.97p, 25p, 24p, 23.98p

O●

O●

12

30p, 29.97p, 25p, 24p, 23.98p

O●O●

ST 2082-10 M1 ST 425-5 (ST 2036-1)

12G-2SI (IV)3840 x 2160

ST 2082-10 M1 ST 425-5 (ST 2048-1)

12G-2SI (IV)4096 x 2160

4:4:4 (YCbCr/RGB)
4:4:4:4 (YCbCrA/RGBA)
4:4:4 (YCbCr/RGB)
4:4:4:4 (YCbCrA/RGBA)

4:2:2 (YCbCr)
4:2:2:4 (YCbCrA)
4:2:2 (YCbCr)
4:2:2:4 (YCbCrA)

KEY
O - Optional
O● - Optional Generator with PHQXLO-GEN / PHQXPO-GEN option and Analyzer† Note: Optional HDR formats require
PHQXLO-HDR / PHQXPO-HDR

www.leaderphabrix.com

27

Supported 4K/UHD IP Formats
The following 4K/UHD ST 2110-20 formats are optional with:
PHQXLO-IP-25G and PHQXLO-UHD or
PHQXPO-IP-25G and PHQXPO-UHD.
Resolution

Sampling
Structure

Pixel
Depth

Frame/Field Rate

2110
HDR†

2110
SDR

3840 x 2160

4:2:2 (YCbCr)

860p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

A

3840 x 2160

4:2:2 (YCbCr)

1060p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

O●●3840 x 2160

4:2:2 (YCbCr)

1260p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

O●●3840 x 2160

4:4:4(YCbCr/RGB)

830p, 29.97p, 25p, 24p, 23.98p

OA

A

3840 x 2160

4:4:4(YCbCr/RGB)

1030p, 29.97p, 25p, 24p, 23.98p

O●●3840 x 2160

4:4:4(YCbCr/RGB)

1230p, 29.97p, 25p, 24p, 23.98p

O●●4096 x 2160

4:2:2(YCbCr)

860p, 59.94p, 50p, 48p, 47.97p, 30p, 29.97p, 25p, 24p, 23.98p

OA

A4096 x 2160

4:2:2 (YCbCr)

1060p, 59.94p, 50p, 48p, 47.95p , 30p, 29.97p, 25p, 24p, 23.98p

O●●4096 x 2160

4:2:2 (YCbCr)

1260p, 59.94p, 50p, 48p, 47.95p , 30p, 29.97p, 25p, 24p, 23.98p

O●●4096 x 2160

4:4:4(YCbCr/RGB)

830p, 29.97p, 25p, 24p, 23.98p

OA

A4096 x 2160

4:4:4(YCbCr/RGB)

1030p, 29.97p, 25p, 24p, 23.98p

O●●4096 x 2160

4:4:4(YCbCr/RGB)

1230p, 29.97p, 25p, 24p, 23.98p

O●●

The following 4K/UHD ST 2110-20 extended formats are optional with:
PHQXLO-IP-25G, PHQXLO-UHD and PHQXLO-EUHD or
PHQXPO-IP-25G, PHQXPO-UHD and PHQXPO-EUHD.
Resolution

Sampling
Structure

Pixel
Depth

Frame/Field Rate

2110
HDR†

2110
SDR

3840 x 2160

RGB:444860p, 59.94p, 50p, 48p, 47.97p

OA

OA

3840 x 2160

RGB:4441060p, 59.94p, 50p, 48p, 47.97p

O●O●

3840 x 2160

RGB:44412

60p, 59.94p, 50p, 48p, 47.97p

O●O●

3840 x 2160

YCbCr:444

860p, 59.94p, 50p, 48p, 47.97p

OA

OA

3840 x 2160

YCbCR:444

1060p, 59.94p, 50p, 48p, 47.97p

O●O●

3840 x 2160

YCbCR:444

1260p, 59.94p, 50p, 48p, 47.97p

O●O●4096 x 2160

RGB:444860p, 59.94p, 50p, 48p, 47.97p

OA

OA

4096 x 2160

RGB:4441060p, 59.94p, 50p, 48p, 47.97p

O●O●4096 x 2160

RGB:44412

60p, 59.94p, 50p, 48p, 47.97p

O●O●4096 x 2160

YCbCr:444

860p, 59.94p, 50p, 48p, 47.97p

OA

OA

4096 x 2160

YCbCR:444

1060p, 59.94p, 50p, 48p, 47.97p

O●O●4096 x 2160

YCbCR:444

1260p, 59.94p, 50p, 48p, 47.97p

O●O●

4K Formats

KEY● - Generator with PHQXLO-GEN / PHQXPO-GEN option and Analyzer
O - Optional
O● - Optional Generator with PHQXLO-GEN / PHQXPO-GEN option and Analyzer
A - Analyzer Only
OA - Optional Analyzer† Note: Optional HDR formats require PHQXLO-HDR / PHQXPO-HDR

28

QxL / QxPQxL Dimensions and Installation
Desktop

Single Rack mount tray
with cover
PHQXK1

Dual Rack mountPHQXK2QxP Dimensions

215mm
with Desktop Kit
211mm without
Desktop kit

3U 19-inch rackmount kit
PHQXK7 - 1 x QxP Chassis
PHQXK8 - 2 x QxP Chassis

310mm

166mm

132mm

148mm

215mm
with Desktop Kit
211mm without
Desktop kit

338mm

www.leaderphabrix.com

29

Notes:30

QxL / QxPNotes:www.leaderphabrix.com

31

www.leaderphabrix.com

This brochure is to be used for informational use only and is subject to change without notice and should not be construed as commitment
by Leader Electronics of Europe Limited. Leader Electronics of Europe Limited assumes no responsibility or liability for errors or
inaccuracies that may appear in this brochure. Please visit www.leaderphabrix.com for latest product information.
August 2024
E&O - As our policy is one of continuous improvement, we reserve
the right to vary details without prior notice.