HIGH POWER LASER BEAM PROFILER / NFP MEASUREMENT & ANALYSIS OF HIGH POWER LASER
M-Scope type HS

Synos

SOPHISTICATED OPTICAL BEAM NFP MEASUREMENT OPTICS FOR HIGH POWER LASER

Optical beam proﬁle measurement optics, customized especially for output ~10W class high power laser.

M-Scope type HS is optics for beam proﬁle measurement of output ~10W class high power
lasers. After passing through the objective lens, the luminous ﬂux from sample is 99.99%
attenuated by two-stage beam sampler, and imaged on the detector.
【Features】○Attenuation of incident beam with two-stage beam sampler and ND ﬁlters○Various objective lens can be selected (M-Plan Apo NUV/NIR series objective lens)○High-performance NFP measurement system can be constructed by using Synosʼ optical beam
analysis module AP013 together.
【Optics selection】* Please contact us regarding the measurement wavelength.
○for 850-940nm
M-Scope type HS/NIR○for 400-450nm
M-Scope type HS/BL【Available detector】【Summary of speciﬁcation】○Hi-resolution CMOS detector ISA071/ISA071GL○Measurement method: dedicated optics & image processing○Attenuation method: Approx. 99.99% attenuated by two-stage【Option】
beam sampler, and ND ﬁlter (combined) ○Option for M-Scope type HS optics●2× intermediate lens port MS-OP016-RL2○Polarization dependent compensation: Compensated by 2-stage
Intermediate lens unit that doubles the overall magniﬁcation
orthogonal arrangement of attenuationof the optical system. (up to 200× with 100× objective lens）
mirrors in beam sampler●1/2× intermediate lens port MS-OP016-RLH○Target input power:
Approx. ~10W
Intermediate lens unit that halves the overall magniﬁcation○Objective lens:
M-Plan Apo NUV, M-Plan Apo NIR
of the optical system.
○Objective lens change: By manual revolver●Coaxial epi-illumination port MS-OP016-CEP○Intermediate lens:
1×
Coaxial epi-illumination port with removable half mirror.
○Epi-illumination:
Option●Dummy ﬁlters MS-OP016-DF○Camera mountC mount
wedge type dummy ﬁlters for pulse/low power measurement【Standard component】〇Accessories
for optics○Main optics:
1●Objective lens, ND ﬁlter, coaxial epi-illumination light source,○Optics base:1
optics bench, etc.
M-Scope type HL

HIGH POWER LASER NFP MEASUREMENT OPTICS

Optical beam proﬁle measurement optics, customized especially for high power laser.

M-Scope type HL is optimized especially for optical beam proﬁle measurement of high power
laser. Approximately 5% of the optical beam emitted from sample is reﬂected by beam sampler
which is installed in front of objective lens. Reﬂected beam is introduced to imaging detector
through NFP optics.
【Features】○Attenuation of incident beam with beam sampler before objective lens, and ND ﬁlter○High-performance NFP measurement system can be constructed by using Synosʼ optical beam
analysis module AP013 together.
○Optical magniﬁcation is maximum 20x (option, 10x objective lens and 2x intermediate lens.）【Summary of speciﬁcation】○Measurement method:dedicated optics & image processing○Attenuation method: Approx. 95% attenuated by beam sampler, 【Available detector】○Hi-resolution CMOS detector ISA071/ISA071GL
and ND ﬁlter (combined)○Measurement wavelength:Select one wavelength from the【Option】
range of 400 nm to 1100 nm○Option for M-Scope type HL optics○Target input power: Approx. ～ 10W●2× intermediate lens port MS-OP011-RL2○Objective lens:M-plan 10x/N.A.0.28
Intermediate lens unit that doubles the overall magniﬁcation○Intermediate lens: 1×of the optical system. (up to 20× with 10× objective lens）○Field of view:
Approx. 706×529μm●1/2× intermediate lens port MS-OP011-RLH○Pixels resolution:
Approx. 0.345μm
Intermediate lens unit that halves the overall magniﬁcation* The value of ﬁeld of view and pixel resolution is approximate value whenusing Hi-resolution CMOS detector ISA071/ISA071GL.

○Epi-illumination:Option○Camera mount:
C mount【Standard component】○Main optics:
1○Optics base:1
Coaxial epi-illumination

Objectivelens

Beam sampler

Beam damper

https://www.synos.jp

Lens forillumination
Imaging detectorof the optical system.
●Coaxial epi-illumination port MS-OP011-CEP
Coaxial epi-illumination port with removable half mirror.
○Accessories for optics●Objective lens, ND ﬁlter, coaxial epi-illumination light source,optics bench, etc.

☞Technical information 【Simple structure of M-Scope type HS】
The light ﬂux emitted from the sample is attenuated to approximately
99.99% by two beam samplers installed in the latter stage of the objective
lens. The beam reﬂected by the beam sampler is absorbed by the beam
damper installed in the optical system. The beam that has passed throughthe beam sampler is further attenuated to an appropriate amount by ND
ﬁlter and then introduced to image detector for image processing analysis.

Focusing lens

13

SYNERGY OPTOSYSTEMS CO., LTD