air.IQ

Moisture analyzer
packaged solution
Features

Applications

air.IQ simplifies the selection and installation of your moisture
analyzer. Install the moisture probe, wire your power andoutputs to the terminal strip, and connect your gas to theinlet fitting.

The standard air.IQ package is designed for moisturemeasurement in any inert gas application, in industrial
environments classified as safe areas, where the process
gas pressure is slightly positive to a maximum of 200 psig.
It combines the Panametrics dew.IQ and IQ.probe with 50
years of sample system design, to deliver the moisturemeasurement you have come to trust.

• Wall mounted NEMA 4X package• Includes the analyzer display, moisture probe,
interconnecting cable, and sample system• Features the dew.IQ moisture analyzer• The IQ.probe makes installation and start-up easy
• Sample system provides isolation, filtration, pressureand flow indication, pre-wired, and a clear door for easy
viewing of all readings

Markets and applications served include:• Industrial gas• Air dryer/clean dry air• Plastics drying• Pharmaceutical• Aerospace• Power generationOrdering configuration
air.IQ is comprised of the following items:• DEW.IQ-3-6-1-0• IQ.PROBE-2-W-0-0-0-0• 733-1155-00

Application parameters• Inert gases such as air, nitrogen, SF6• Sample gas pressure: 0 to 200 psig
• Sample gas temperature: 0 to +50 C• Moisture content: -110 to +20 C dew/frost point,
non-condensing• Power requirements: 100 - 240 VAC @ 50 - 60 Hz

dew.IQ specifications*

European certification
Complies with EMC directive 2004/108/EC and 2006/95/
EC low voltage directive (installation category II, pollution
degree II)
Input
Moisture signal from an M series probe or IQ.probe
Analog output
Single internal isolated recorder output, internally opticallyisolated, 10-bit (0.1%) resolution
Switch-selectable outputs• 0 to 2 V, 10k Ω minimum load resistance• 0 to 20 mA, 400 Ω maximum series resistance• 4 to 20 mA, 400 Ω maximum series resistance• User-programmable within the range of the instrument
and the corresponding sensor or transmitter
Alarm relays• One fail-safe fault relay• Two standard form C relays SPDT, rated for 3 A at 250
VAC/30 VDC• Set to any level within the range of the instrument;
programmable from the front panel
Alarm set point repeatability±0.2°F (±0.1°C) dew point
Datalogger
32 GB capacity with MicroSD card, 2 GB card included
Display
128 x 64 matrix LCD* Refer to dew.IQ and IQ.probe data sheets for complete specification details

Display functions
Dew point temperature in °F or °C, ppmv with a constantpressure input, or sensor signals for diagnostics
Power requirements
Universal power 100-240 VAC @ 50-60 Hz
24VDC option available

Temperature• Operating: –20° to 60°C (–4° to 140°F)• Storage: –40° to 70°C (–40° to 158°F)

Warm-up time
Meets specified accuracy within three minutes

IQ.probe specifications*Sensor typeThin-film aluminum oxide

Dew/frost point temperature• Overall range capability: –110° to 60°C (–166° to 140°F• Standard: –80° to 20°C (–112° to 68°F) with datato –110°C (–166°F)
Calibrated accuracy at 77°F (25°C)• ±3.6°F (±2°C) above -148°F (–100°C)• ±5.4°F (±3°C) below -148°F (-100°C)

Repeatability• ±0.4°F (±0.2°C) above -148°F (–100°C)• ±0.9°F (±0.5°C) below -148°F (–100°C)

Start-up procedure

Shut-down procedure• Insert moisture probe into the sample cell

• Slowly close the inlet needle valve• Start with the inlet valve and the valve on
rotameter fully closed• Slowly open the valve on the rotameter until the pressure
on the pressure gauge is 0 psig

• For dew points at process pressure, slowly open the inletvalve until fully open; then crack the valve on the rotameterto get flow on scale• Remove the moisture probe• For dew points at atmospheric pressure, fully open thevalve on the rotameter; then crack the inlet needle valve
on the rotameter to get flow on scale!

100-240 VAC
50-60 Hz, 5W

1L12L2N

3456784ALARM
A

FAULT
RM

10 11 12 13 14 15

9 10 11 12 13 14 15

ALARM
B

- NC C NO NC C NO NC C NO

7 8 96ANALOG
PUT

5+4ALARM
B

100-240 VAC
50-60 Hz, 5W

ANALOGOUTPUT

L1

L2N

ALARM
A

FAULT
ALARM

10 11 12 13 14 15- NC C NO NC C NO NC C NO

7 8 965+

COPYRIGHT 2020
ALL RIGHTS RESERVED

BAKER HUGHES CO.

CONDUIT
HUB

CONDUIT
HUB!

23100-240VAC
CUSTOMER
SUPPLY

L1 L2 N

14789 10 11 12 13 14 15

ALARM
B

ALARM
A

FAULT
ALARM

- NC C NO NC C NO NC C NO

6ANALOGOUTPUT+5IQ.probe

dew.IQ CONNECTOR PROBE WIRING

BILL OF MATERIALS
Shannon - Ireland
Billerica - MADRAWN

APPROVED

J.R. 25-OCT-12

T.K. 25-OCT-12

ECM

BMSAMPLE SYSTEM BILL OF MATERIAL

500000348343

CHECKED

INTERNAL REF.

PART NO.

1UB14750-2337

2035255-184

DCOMMENTSair.IQ-1-1 AC VERSION
1/4" IMPERIAL FITTINGS

E.J. 25-OCT-12
DWG
ITEM

REVBM733-1155-00

DESCRIPTION

MFG.

MFG. PART No.

QUANTITY

UNION BULKHEAD ASSY, 316SS, 1/4" COMPRESSION, WITH SELF-SEALING WASHER NEMA TYPE
4X & 6, IP66/68

SWAGELOK CO.

SS-400-61

2SS-1RS41VALVE, NEEDLE, 5000 PSI (344 BARG), 316SS, 1/4" COMPRESSION ENDS

SWAGELOK CO.

BAKER HUGHES

3750-2123-00

AIR.IQ SAMPLE CELL ASSEMBLY, 5000 PSIG RATED, 316SS, INCLUDES FILTER ELEMENT, PVDF
FLOUROCARBON, +99.5% EFFICIENT AT REMOVAL OF 0.1 MICRON PARTICLES & AEROSOLS, 1/4"COMPRESSION FITTINGS - 1 X MALE CONNECTOR/1 X MALE ELBOW

4146M5241

IQ PROBE, 3/4"-16 PROBE MOUNT, W SHIELD (SS), STANDARD CALIBRATION -110° TO +20°C DATAPANAMETRICS

IQ.PROBE-2-W-0-0-0-0-0

1

SWAGELOK CO.

PGI-40M-PG200-CANX

1KING INSTRUMENT CO

74C214G021223910

115136D

443-069-10

GAUGE, PRESSURE, 1-1/2" (40MM), 316 SS, ± 2.5 % of span (ASME B40.1 Grade C, EN 837-1 Class 2.5,JIS B7505 Class 2.5), AMBIENT OPERATING TEMP. –40 to 140°F (–40 to 60°C), 1/8" NPTM CENTREBACK MOUNT, RANGE 0 to 200 PSI/BAR

6098B

750-414

FLOWMETER ASSY, BOROSILICATE GLASS METERING TUBE, 200 PSIG RATED, COMPLETE WITH316L SS INTEGRAL INLET FLOW CONTROL VALVE, FLOW RANGE 1.9 to 19.5 SCFH / 54 to 540 SLPH,CAL AIR @ STP, 316L SS FITTINGS, SAPPHIRE FLOAT AND VITON 'O' RING, AMBIENT TEMP 33°F to125°F (1°C to 52°C), 2 x 1/4" MALE ELBOWS

7146M5237

DEW.IQ MOISTURE ANALYSER, PANEL MOUNT PACKAGE, 100-250VAC, STANDARD PROGRAM

PANAMETRICS

DEW.IQ-3-6-1-0

18418-200

MTS6 / DEW IQ MOUNTING BRACKETDIMARK SHEETORDER BY DESC

1750-2353

ENCLOSURE ASSY, FIBREGLASS WITH WINDOW, HINGED COVER, 14"x12"x8", NEMA/EEMAC Type4, 4X, 12, 13 CSA File No. 42186: Type 4, 4X, 12, 13, IEC 60529, IP66, UL 508A LISTED, INCLUDES
MOUNTING PLATE RAL6033 AND BREATHER GLAND (NEMA 4X & IP66 RATED)HOFFMANENGINEERING CO.

A14128CHQRFGW

1213-2000

BARRIER BLOCK, 15 POS., INSULATION MATERIAL: MOLDED MONOBLOCK, GENERAL PURPOSEPHENOLIC, BLACK, UL RATED 94V-1, OPERATING TEMP -55° TO +300°F, VOLTAGE RATING 250 VAC
MAX., CURRENT RATING 15 AMPS MAX.

CINCH15-140

19H185Q

1011213-2001

TERMINAL BLOCK MARKER, 1 TO 15, FOR 140 SERIES BARRIER BLOCK

12421-2520

TERMINAL COVER FOR 15 PIN TERMINAL BLOCK, FR4 GLASS EPOXY 0.031" THICK

13442-1036

LABEL,SAMPLE SYSTEM,MTS5,OUTPUTS

14442-1345

LABEL. SAMPLE SYSTEM. MTS6 POWER STRIP

115442-1355

SUB COMPONENT LABEL

116148M2218SPACER UNTHREADED SS 13MM DIA.x22MM LENGTH FOR M6 BOLT17410-2006

GREEN VINYL CAP P/N VC-562-81819230-098

CG1BNP

419-087

CINCHMS-15-140

1EVANS MACHINE CO.,INC.

FABRICATED

11McMASTER-CARR

92871A352

22CABLE, 4 x 26 AWG CONDUCTORS + SHIELD (BROWN/WHITE/BLUE/BLACK), M8 CONNECTOR,24VDC RATED, BLACK OUTER SHEATH Ø4.2, 0.3 METRE LENGTH
CABLE GLAND, UNARMOURED, BRASS, NICKEL PLATED - 1/2"NPT SUITABLE FOR CABLE WITHOUTER DIA 4 - 8.4MM, ATEX: II 2 GD Ex d IIC / Ex e II (BAS 01ATEX1272X & SIRA 09ATEX1221X),
IECEx : Ex d IIC / Ex e II / Ex tD A21 (SIR 07.0096X), NEC CL.I ZONE 1 (CSA 2627370), EAC Exd IICU(RU C-GB.ГБ06.В.00098)

TURCK INC.

PKG 6M-0.3/CS10922

1PEPPERSA2LBF/NP/16/050NPT

220419-943

CONDUIT HUB, MATERIAL : ZINC, CONDUIT CONNECTION: 1/2" NPTF, UL LISTED/CSA CERTIFIEDMcMASTER-CARR

7513K81221255-440

TUBING, 316/316L SEAMLESS STAINLESS STEEL, 1/4" X .035" WALL, NACE COMPLIANT, MATERIALCERTIFICATION TO EN10204 3.1

SWAGELOK CO.

SS-T4-S-035-20

0.222750-2357-00

CABLE ASSEMBLY FOR AIR.IQ SIGNALS / VAC POWER, INCLUDING EARTHING

BAKER HUGHESREVABCDECM NO.
08805
09129
500000284659
500000348343

APPD/DATE
10-SEP-2013
24-OCT-2014
13-MAR-2018
13-MAY-2020

REV

ECM NO.

APPD/DATEREVECM NO.

APPD/DATE

1NOTES
REF. DRAWING : 733-1155

Panametrics, a Baker Hughes Business, provides solutions inthe toughest applications and environments for moisture,oxygen, liquid and gas flow measurement. Experts in flaremanagement, Panametrics technology also reduces flareemissions and optimizes performance.
With a reach that extends across the globe, Panametrics’
critical measurement solutions and flare emissions
management are enabling customers to drive efficiency
and achieve carbon reduction targets across criticalindustries including: Oil & Gas; Energy; Healthcare; Water andWastewater; Chemical Processing; Food & Beverage andmany others.
Join the conversation and follow us on LinkedIN
linkedin.com/company/panametricscompany

Copyright 2024 Baker Hughes Company. All rights reserved.
BHCS38743(05/2024)

panametrics.com