SAF series

SPEC.SHEET

100Hz Maximum

Low Frequency Transmitter

Model：SAFU(With indication function & sensor power)
Output Specifications

Model

SAFU－2□－□Input
0 : Open collector
1 : Voltage pulse
2 : Line driver

When the output range lower limit is zero, (even if zeroadjustment results in a negative value), the output value will
not be negative.
DC Current

4 to 20mA DC

Allowable
loadresistance
700 or less0 to 20mA DC

700or less0 to 5%95 to 105%

0 to 12mA DC

1.2kor less0 to 5%95 to 105%

0 to 10mA DC

1.2kor less0 to 5%95 to 105%

1 to 5mA DC

2.4kor less

-5 to 5%95 to 105%

Zeroadjustment
range
0 to 5%Span
adjustment
range
95 to 105%

OutputPower supply
0 : 100 to 240V AC
1 : 24V AC/DC

Zeroadjustment
range-5 to 5%Span
adjustment
range
95 to 105%

DC Voltage

0 to 1V DC

Allowable
loadresistance
100 or more

0 to 5V DC

500

or more0 to 5%95 to 105%

1 to 5V DC

500

or more-5 to 5%95 to 105%

0 to 10V DC

1kor more

0 to 5%95 to 105%

Output

How to orderSpecify a model & frequency.
(e.g.) SAFU-20-0 (Frequency 50Hz)
Default value (If not specified, shipped as the following default value)
Input frequency
100Hz
Output4 to 20mA DC

Input specifications
Open collector
Frequency Range: 0.001 to 1Hz
0.001 to 100Hz
Minimum pulse width: 4 s or more (for ON and OFF)
Input detection voltage/current: ON: Max. 30mA (30V or less)
OFF: Residual voltage 0.5V orlessAction input conditions: ON: 200 or less
OFF: 100k or more
Voltage pulse
Frequency range
Minimum pulse width
Waveform
Detection level
Input impedanceInput amplitude: 0.001 to 1Hz
0.001 to 100Hz: 4 s or more (for High and Low)
: Rectangular, sine waveform orsimilar: Low: 1V DC or less
High: 2V DC or more
: 10k or more
: 2 to 50Vp-p

Line Driver
AM26LS31 or equivalent
Receiver: AM26LS32 or equivalent

Performance
Reference accuracy
Output accuracy
Indicating accuracy
Response time
Temperature coefficientInsulation resistance: Within 0.1%
: Within 0.1%
: Within Reference accuracy 1 digit: 200ms + Pulse cycle: 0.015%/: 10M or more, at 500V DC(Input - Output - Power)Dielectric strength: 2.0kV AC for 1 minute(Input - Output - Power)
Frequency sampling period: 500ms

General structure
Case: Flame-resistant resin Color: Light gray
Front panel : Membrane sheet
Setting: By the front keypad
Indication : Input display:
7-segment, Red LED display 4-digit
Character size, 7.4 x 4.0mm (H x W)
Output display:
7-segment, Green LED display 4-digit
Character size, 7.4 x 4.0mm (H x W)
Power indicator: Green LED

Power indicator
Input display
Output display
UP keyDOWN key

MODE key
SUB MODE key

SAFUInstallation specifications
Power supply: 100 to 240V AC, 50/60Hz
24V AC/DC, 50/60Hz
Allowable voltage range: 85 to 264V AC
20 to 28V AC/DCPower consumption: Approx. 9VA
Power supply for sensor: 12V DC 5%, 25mA
Ambient temperature : -5 to 55
Ambient humidity: 35 to 85%RH (non-condensing)
Weight: Approx 120g
Mounting: DIN rail mounting
External dimensions: 22.5 (W) x 75 (H) x 100 (D)mm

FerrulesTerminals from 1 to 4
Insulation sleeve attached (Phoenix Contact GMBH & CO.)AI0.25-8YE
0.2－0.25mm2AI0.34-8TQ0.25－0.34mm2
AI0.5-8WH
0.34－0.5mm2
AI0.75-8GY
0.5－0.75mm2
AI1.0-8RD
0.75－1.0mm2
AI1.5-8BK
1.0－1.5mm2
Crimping pliers (Phoenix Contact GMBH & CO.)CRIMPFOX ZA3CRIMPFOX UD6◆Attached functions
Power failure countermeasure:The data is backed up in non-volatile IC memory.
Self diagnosis:
The CPU is monitored by a watchdog timer, and whenan abnormal status is found on the CPU, the unit isswitched to warm-up status after turning all outputs off.
Detecting unconnected sensor:
If pulse is not detected for a constant period (100sec),
the unit will revert to the initial status (0Hz).

Terminals from 5 to 9
Insulation sleeve attached (Phoenix Contact GMBH & CO.)AI0.25-8YE
0.2－0.25mm2
AI0.34-8TQ
0.25－0.34mm2
AI0.5-8WH
0.34－0.5mm2
Crimping pliers (Phoenix Contact GMBH & CO.)CRIMPFOX ZA3CRIMPFOX UD6

Circuit configuration and Terminal arrangement
Settings
Function keys
(1) UP key: Increases the numeric value.
(2) DOWN key: Decreases the numeric value.
(3) MODE key: Switches the setting mode.
(4) SUB MODE key : Press with the MODE key toproceed to the Setup mode.
Setting items
Setting by pressing the MODE key for 3 seconds(1) Output zero adjustment(2) Output span adjustment
Setting by the MODE key and SUB MODE key(1) Set value lock(2) Frequency high limit value(3) Output 0% value(4) Output 100% value(5) Decimal point place(6) Output type(7) Output Normal/Reverse(8) Shutdown threshold value(9) Output volume during shutdown(10) Display selection(11) Indication time+InputPower supplyfor sensor+

Insulation
circuit5InputcircuitOutputcircuit

3Powercircuit

167 Fixed voltage Display/Setting
8circuit 12V

key circuit

942+
Output+Power supply-

Input connection example
+

567-

12V

567812V

899Open collector

Voltage pulse

5Line driver

678

Displays and indicators
Input display: Indicates the input value.
Indication of 10000 or more: The lower 4 digits flash.
0 flashes when pulse is absent.
When input frequency is lower than shutdown threshold value,
the input value flashes.
Over range: “” flashes on the Input display.
(1.1 times frequency high limit value)
Warm-up indication: For approx. 2 seconds after power-on, theinput type is indicated on the Input display,
and the output type is indicated on theOutput display.
Output display: Indicates the output volume in a percentage
form (%).
Power indicator: The green LED lights when power-on.

CPU9

External Dimensions (Scale: mm)

SHINKO TECHNOS CO., LTD.

Tel: 81-72-727-6100
URL: http://www.shinko-technos.co.jp/e/ E-mail: overseas@shinko-technos.co.jp

20090605