.............................................................................................................................................................

TOX®-Pressing Monitors
Data sheet 100.10
2020 / 02TOX® PRESSOTECHNIK GmbH & Co. KG, Riedstrasse 4, 88250 Weingarten / Germany
Find your local contact at: www.tox.com
1

TOX®-Pressing Monitors.............................................................................................................................................................

Real-time monitoring of production processes
The TOX®-Pressing Monitors monitor processes where aprecisely defined functional correlation between forces anddistances must be examined. This equipment is used widely to monitor joining, riveting, pressfitting and caulking
processes and leads to continous improvement of quality
in production.
Here, during measuring the devices read force-path datapairs belonging together from two measuring channels‚X‘ and ‚Y‘. The data are stored in a memory and can bedisplayed graphically. The resulting force-path function iscompared with specified maximum ratings of the configured window values or envelope. If the maximum ratings arecomplied with, an IO message is issued, otherwise an NIO
message.

Configuration
Thanks to the built-in display and the understandable menu
the device can be comfortably configurated and parameterized. The complete production-data is stored and can beedited and displayed easily.
The change of piece parts is quickly performed due to 64
specific parameter settings, which can be selected either
directly by the touch screen or by PLC.

Features Real-time monitoring of production processes
 Provides important data about process sequence Special monitoring by using window technique andenvelopes
 Individual measuring programs can be selected User interface language can be switched Backup of measurement data and final values
 Various password levels
 Colour touchscreen

The EPW 400 monitors pressing processes and is available
as built-in device and wall mounted version.

EPW 500FP: the economical alternative is available as wall
mounted version.

Extensions
 TOX®-Evaluation Box with extended display andacceptance function
 Fieldbus interfaces

Compact design
 Swivelling housing
 Compatible with Rittal CP60 and VESA 75 x 75 mm fixtures

2

www.tox.com

TB 100.10_202002.en

.............................................................................................................................................................

Technical data

EPW 400

EPW 500FP

Application

Monitoringof pressing processes

Monitoring of pressing processes, particularly forthe bench top presses TOX®-FinePress

Supply voltage

24 V DC

24 V DC

Current consumption

≤1A

≤1A

Display

5.7" colour touchscreen

7" colour touchscreen

TFT LCD 640 x 480 (VGA)
Colour depth 16 bit

LCD 800 x 480 (WVGA)
Colour depth 16 bit

Inputs / Outputs- Digital: 16 inputs / 8 outputs- Analogue: 2 outputs- Digital: 13 inputs / 8 outputs- Analogue: 3 outputs

Number of measuring programs

64

128

Dimensions W x H x D (mm)Built-in version: 210 x 185 x 123.5(D 99 for two slots)
Wall mounted version: 320 x 305 x 149

Housing dimensions: 244 x 207 x 75

WeightBuilt-in version: 1.5 kg
Wall mounted version: 4 kg

2.5 kgInterfaces connections

Ethernet, USB, Digital In/OutUSB, Digital In/Out

Fieldbus interfaces

Ethernet, PROFIBUS DP, PROFINET CU, DeviceNet,
Ethernet IP, CANopen, PROFINET LWL, EtherCA–Protection class

Built-in device
Wall mounted version

IP54 front panel
IP20 housing
IP54

IP 67 front panel
IP 40 rear of housing

Measuring channels:
Force DMS
Force standard signal
Path standard signal (1.1 mV / V)

Measuring channel resolution

16 bit16 bitMax. sampling rate

2000 Hz2000 Hz

Monitoring by means of window and / or envelope
technology

max. 10 windows

max. 10 windowsPassword protection

Display of last measured curves in diagram

1010

Number of last final values in memory
Number of last measured curves in memory

1000
1001000
100Backup

100 measured curves and 1000 final values transferrable to USB memory stick as CSV file

100 measured curves and 1000 final values transferrable to USB memory stick as CSV file

Backup with TOX®softWare

TB 100.10_202002.en

3

TOX®-Pressing Monitor EPW 400.............................................................................................................................................................

Connections and Dimensions
Connections wall mounted version
X1

OFFY1

ONI/O

2
3

5X4-X1

X3-X2

4111.3

1

Connections build-in version

29451 Digital input / output
2 Fieldbus interface (option)
3 Reserve
4 Travel transducer X15 Force transducer Y1

67986 Main switch
7 Supply voltage 24 V DC
8 USB interface
9 Ethernet interface

1

7

8Dimensions wall mounted version

282

290

149

292
305

320

140

Dimensions built-in version
200

1853

10

20

1010

3

V-Bus ETH
USB
USB

150

82.5

DIG.-E/A

175

CAN
123.5

4

www.tox.com

TB 100.10_202002.en

24.5

TOX®-Pressing Monitor EPW 500FP.............................................................................................................................................................

Connections and Dimensions
Connections
1726354

1 DMS force signal
2 Standard signal force
3 Analog output
4 USB interface
5 Power supply
6 Digital input / output
7 Travel sensor potentiometricand with standard signal output

Dimensions

50907075

207155244
215

Subject to technical modifications.

TB 100.10_202002.en

5