.............................................................................................................................................................

TOX®-Process Monitoring
CEP 400T
Data sheet 100.20
2024 / 01

TOX® PRESSOTECHNIK GmbH & Co. KG, Riedstrasse 4, 88250 Weingarten / Germany
Find your local contact at: www.tox.com
1

TOX®-Process Monitoring CEP 400T.............................................................................................................................................................

The TOX®-Process monitoring monitors clinching and joining processes and provides important data about the process sequence.
Functions: Real names for program memory, up to 40 characters 64 sensor configurations per channel
 4 password levels: operator, setup technician, integrator,
manufacturer
 16 digital inputs and 8 digital outputs Status indicator menu of digital I/O ports
 Stores up to 1000 values with time stamp Includes TOX®softWare for parameterization and datarecording Choice of panel mount unit or wall mount unit IP54
 Ethernet interface
 Provides important information about the process
sequence
 Special monitoring by means of window technology forclinching processes Individual measuring programs available
 Storage of final values
 Several password levels
 Color touchscreen Versions with 1 / 2 / 4 / 8 / 12 channels
 Individual evaluation with 1-/2 channel versions

2

www.tox.com

TB 100.20_202401.en

Optional expansion:
 Field bus interfaces of all common bus systems*
 Process sequence control for TOX®-Clinching Processes
 Service modules such as- Remote maintenance- Test control unit to check measuring chain
 Analog output 0 – 10 V
 Evaluation unit for bench top presses TOX®-FinePress
 Display of control dimension X (absolute value)*With field bus connection, individual evaluation IO / NIO andend values are displayed, independent of the number ofmeasuring channels in the device.

.............................................................................................................................................................

Technical data

CEP 400T

Application

Monitoring of force processes

Supply voltage

24 V DC

Current consumption≤1ADisplay

5.7" color touchscreen
TFT LCD 640 x 480 (VGA)
Color depth 16 bit

Inputs / Outputs

Digital: 16 inputs / 8 outputs

Number of measuring programs

64

Dimensions W x H x D (mm)Built-in unit: 210 x 185 x 123.5(D 99 for two slots)
Wall mounted unit: 320 x 305 x 149 (1 / 2 channel)
404 x 360.5 x 161 (4 / 8 / 12 channel)

WeightBuilt-in unit: 1.5 kg
Wall mounted unit: 4 kg

Interfaces

Ethernet, USB, Digital In/Out

Fieldbus interfaces

PROFIBUS DP, PROFINET CU, DeviceNet, Ethernet IP, CANopen, PROFINET LWL, EtherCAT
separate gateway for 12 channel devices

Protection class

Built-in unit

IP54 front panel
IP20 housing
IP54Wall mounted unit
Number of measuring channels

1 / 2 / 4 / 8 / 12

Measuring channels:
Force DMS
Force standard signal

Measuring channel resolution

12 / 16 Bit

Max. measuring interval

2000 Hz

Monitoring by means of window technology

Password protection

Number of last final values in memory

1000

Data backup

1000 final values transferrable to USB memory stick as CSV file
Data backup with TOX®softWare

TB 100.20_202401.en

3

TOX®-Process Monitoring CEP 400T.............................................................................................................................................................

Connections
Wall mounted unit
OFFI/O

23X4111.3

1

ON-X1X3-X2

45

1 Input / Output digital
2 Fieldbus interface (option)
3 Reserve
4 XK 1-4 Force transducer
5 XK 5-8 Force transducer
6 XK 9-12 Force transducer

6871097 Main switch
8 Supply voltage 24 V DC
9 USB interface
10 Ethernet interface

Build-in unit
6542
101

4

www.tox.com

9TB 100.20_202401.en

7

TOX®-Process Monitoring CEP 400T.............................................................................................................................................................

Dimensions
Wall mounted unit 1 / 2 channel
320

290

305

292

149

140282

Wall mounted unit 4 / 8 / 12 channel
404

344

360.5

346

161

361.5

153

Built-in unit

200
53

1020

101024.5

183

V-Bus

150

DIG.-E/A

82.5

175

ETH
USB
USB

CAN

123.5

Subject to technical modifications.

TB 100.20_202401.en5