Protec P3000(XL) Helium Leak Detector

Wise Technology

Sniffer Leak Detection You Can Rely On

INFICON Protec P3000(XL) Helium Sniffer Leak Detector is
specifically designed for full-time sniffing applications in demanding
production environments.

1

Protec P3000(XL) Helium Leak Detector

FEATURES

I-Guide operator guiding mode ensures your operator is
testing the right locations with the correct technique.

Improved system design compensates for operator error
reducing the potential for missed leaks.

Multiple alarm functions make sure alarms can not beoverlooked.

Built-in PRO-Check reference leak allows for easy and fastcalibration at the production line at any time.

A small display in the ergonomically-designed probe handle
shows the leak rate, so the operator can concentrate on thesniffing process and monitoring the leak rate at the sametime.

Leak rates can be displayed in refrigerant equivalents froma gas library.

Built-in illumination source of the probe helps precisely
position the sniffer tip

New, low-maintenance sensor yields high reliability and low
cost of ownership.

Automatic standby prevents intake of contaminants into thesniffer probe, thus saving filter and sensor life.

Operating software is available in many languages.

ORDERING INFORMATION
Type

Protec P3000(XL)

Protec P3000 Helium Leak Detector, 115 V

520-002

Protec P3000 Helium Leak Detector, 230 V

520-001

Protec P3000XL Helium L Detector, 115 V

520-004

Protec P3000XL Helium L Detector, 230 V

520-0032

Protec P3000(XL) Helium Leak Detector

SPECIFICATIONS
Type

Protec P3000(XL)

Min. detectable leak rate ProtecP3000

1E-7 mbar l/s

Min. detectable leak rate Protec
P3000XL with 300 sccm

1E-7 mbar l/s

Min. detectable leak rate Protec
P3000XL with 3000 sccm

1E-6 mbar l/s

Measuring scale Protec P3000

5 decades

Measuring scale Protec
P3000XL with 300 sccm

5 decades

Measuring scale Protec
P3000XL with 3000 sccm

4 decades

Sensor response time

450 msResponse time incl. sniffer line

< 0.7 sLeak rate units
Refrigerant equivalent leak rates
Dimensions (width x depth xheight)
Weightmbar l/s; Pa m3/s; ppm
g/a; oz/yr; lb/yr
610 x 265 x 370 mm (24 x 10.4 x 14.6 in)
27 kg (60 lbs)

Gas flow Protec P3000300 sccm

Gas flow Protec P3000XL3000 sccm / 300 sccm

Ambient temperature range

10-45°C (50-113°F)

3Protec P3000(XL) Helium Leak Detector

ACCESSORIES
Protec P3000(XL)
Adapter for custom SL P3000 / E3000

525-005

Adapter for custom SL Protec P3000XL

521-015

cable 5 m, fits to 551-100

551-102

Connecting cable ext. display unit, 0.7m

551-103

Cover test leak opening P3000/E3000

525-007

Display unit, rack mounting

551-101

Display unit, table top

551-100

Holder for SL3000

525-006

Oil / Water Protection Tip for SL3000XL

521-016

SL3000-10, 10m length525-003

SL3000-15 for Ecotec E3000, 15m length525-004

SL3000-3, 3m lengths525-001

SL3000-5, 5m length525-002

SL3000XL-10, 10m lengths

521-013SL3000XL-15, 15m lengths

521-014SL3000XL-3, 3m lengths

521-011SL3000XL-5, 5m lengths

521-012

Sniffer probe FT 312

12214

Sniffer probe FT250XL, 250 mm, flexible

521-022

Sniffer probe FT385

12216

Sniffer probe ST200

12218

Sniffer probe ST385

12215

Sniffer tip FT250

12266

Sniffer tip FT312XL, 120 mm, flexible

521-019

Sniffer tip FT385XL, 385 mm, flexible

521-021

Sniffer tip FT600

12209

Sniffer Tip ST 312

12213

Sniffer tip ST312XL, 120 mm, rigid

521-018

Sniffer tip ST385XL, 385 mm, rigid

521-020

Water protection tip for Protec / Ecotec

12246

4

Protec P3000(XL) Helium Leak Detector

CONSUMABLES
Protec P3000(XL)
05Filter pads (25 pcs.) for SL3000XL

521-023

Helium sniffer leak standard S-TL4

12237

Helium sniffer leak standard S-TL5

12238

Helium sniffer leak standard S-TL6

12239

PRO-CHECK reference leak, Helium

521-001

Spare Filter for Oil/Water Protection Ti521-017

Spare Helium gas cartridge for PRO-Check

521-010

www.inficon.com

reachus@inficon.com

Due to our continuing program of product improvements, specifications are subject to change without notice.
RateWatcher is a trademark of INFICON. All other trademarks are the property of their respective owners.
(2013-07)© 2013 INFICON