SB series

SPEC. SHEET

Potentiometer Transmitter

Model:

SBPFeatures
Simple wiring using a plug-in socket
Compact
3-port insulation (Input-Output-Power)

Model
SBP–□□–□
Input signal
01: Total resistance 100
Output signal
01: 4 to 20mA DC
02: 0 to 20mA DC
03: 0 to 12mA DC
04: 0 to 10mA DC

to 10k

05: 0 to 1V DC
06: 0 to 5V DC
07: 1 to 5V DC
08: 0 to 10V DC

Performance

Power supply
0: 100 to 240V AC
1: 24V DC

Accuracy
Response time

: Within 0.2%
: 1 sec. (090%)(Average 0.5 sec)
Temperature coefficient: 0.015%/
Insulation resistance : 10M or more, at 500V DC(Input-Output-Power)Dielectric strength: 2.0kV AC for 1 minute(Input-Output-Power)

How to orderSpecify a model name.
(e.g.) SBP-0101-0

General specifications

Input specifications
Total resistance : 100 to 10k
Reference voltage: 1.0V DC
Minimum span
: 50% or more of the Total resistance

Output specifications
When the output range lower limit is zero, (even if zeroadjustment results in a negative value), output value will
not indicate a negative value.
DC current
Output range4 to 20mA DC
0 to 20mA DC
0 to 12mA DC
0 to 10mA DC

Allowable load
resistance
600 or less
600 or less
1k or less
1k or less

DC voltage
Output range
0 to 1V DC
0 to 5V DC
1 to 5V DC
0 to 10V DC

Allowable load
resistance
100 or more
500 or more
500 or more
1k or more

Case
: Flame resistant resin, Color: Light gray
Panel
: Polycarbonate
Spring type plug : Polyamide, Color: GreenAdjustment: By the front adjustment button
Zero adjustment:
By pressing the Zero adjustment button for 2 seconds
after setting the input potentiometer to any position of
the MIN side, zero position is automatically registered.
Span adjustment:
By pressing the Span adjustment button for 2 seconds
after setting the input potentiometer to any position of
the MAX side, span position is automatically registered.
Indication:
PWR indicator: Green
Lights when power is turned on.
Flashes in 500ms cycles when an error has occurredin non-volatile memory.
ERR indicator: Red
Flashes in 250ms cycles when input is 110% or more.
Flashes in 500ms cycles when input is -10% or less.
Output status selection:
Output status either Normal or Reverse can be selectedby the DIP switch.
No.1 OFF: Normal, ON: Reverse
Momentary power failure: 30msec
Self diagnosis:
The CPU is monitored by a watchdog timer, and whenan abnormal status is found on the CPU, the unit
restarts with the Reset action.

SBP
Circuit configuration, terminal arrangement

Power indicator
Zero adjustment

Error indicator

Span adjustment

Power supply
Output

+++

6

Power
circuit

5Output
circuitOutput-+4
3

Potentiometer

Input

7

Power supplyMIN.

2MINMAXMAX.

1InputcircuitCPUReferencevoltage

External dimensions (Scale: mm)

DIP switch

DIN rail

Spring type plug

Installation specifications
Power supply• 100 to 240V AC 50/60Hz
Allowable voltage range: 85 to 264V AC
Power consumption: Approx. 3.5VA• 24V DC
Allowable voltage range: 20 to 28V DC
Power consumption: Approx. 3.5W
Ambient temperature : -5 to 55
Ambient humidity: 35 to 85%RH (non-condensing)
Weight: Approx. 80g
Mounting method: DIN rail mounting
Be sure to use fastening plates atboth ends of the unit after the unit
is mounted to the DIN rail.
External dimensions : 17.5 (W) x 75 (H) x 85 (D) mm

Environmental specification
RoHS directive complianceFerrules
Made by Phoenix Contact GMBH & CO.
Insulation sleeve attached
ModelConductor cross section
2AI0.25-6BU
0.2 – 0.25mm
2AI0.34-8TQ
0.25 – 0.34mm
2AI0.5-8WH
0.34 – 0.5mm
2AI0.75-8GY
0.5 – 0.75mm
2AI1-8RD
0.75 – 1.0mm
2AI1.5-8BK
1.0 – 1.5mm
2AI2.5-8BU
1.5 – 2.5mm
Crimping pliers
CRIMPFOX ZA3CRIMPFOX UD6SHINKO TECHNOS CO., LTD.

Tel: +81-72-727-6100
URL: http://www.shinko-technos.co.jp/e/E-mail: overseas@shinko-technos.co.jp

20100624