2700G Series
ReferencePressure Gauges

Technical DataFeatures• Precision pressure measurement from 100 kPa(15 psi) to 70 MPa (10 000 psi)
• Accuracy to 0.02% of full scale• Easy-to-use, rugged construction for reliableperformance• Combine with the 700PTPK or 700HTPK pump kits
for a complete portable pressure testing solution for
up to 4 MPa (600 psi) with the PTP-1 pneumatic
pump and up to 70 MPa (10 000 psi) with theHTP-2 hydraulic pump
• Combine with the P5510, P5513, P5514, or P5515
Comparison Test Pumps for a complete bench toppressure calibration solution• Test port is 1/4 NPT Male. 1/4 BSP and M20 X 1.5
adaptors are included standard• USB communications cable and universal power
supply included standard

Combine the 2700G gauge with the P5510, P5513, P5514 (shownhere) or P5515 Comparison Test Pumps for a complete bench toppressure calibration solution.

The 2700G Reference Pressure Gauges provide
best-in-class measurement performance in a rugged,easy-to-use, economical package. Improved measurement accuracy allows it to be used for a wide varietyof applications. It is ideal for calibrating pressure
measurement devices such as pressure gauges, transmitters, transducers, and switches. In addition, it canbe used as a check standard or to provide process
measurements with data logging.
The 2700G has an easy-to-use interface thatallows you to configure the sampling rate, tare value,
damping, and auto-off time interval as well as reset
the min/max pressure. You can also view the remaining
battery life. Select from 21 different standard engineering units, including bar, in H2O, kPa, MPa, and psi.
In addition, the supervisor menu can be used to set
one custom engineering unit. To ensure against incorrect measurements, each of the above functions canbe locked from use through the supervisor menu.
The 2700G Reference Pressure Gauge can becombined with the Fluke Calibration P5500 series ofcomparison test pumps to make a complete pressure
calibration system. The unique test port design of theP5500 series allows for hand tight connection of the2700G without the use of PTFE tape.

P5510 Pneumatic Comparison Pump
• Pressure range: 0 to 2 MPa (300 psi)
• Vacuum range: 0 to 80 kPa (24 inHg)• Built-in hand pump as pressure/vacuum source• High-quality needle valve for fine control

P5513 Pneumatic Comparator• Pressure range: 0 to 20 MPa (3000 psi)
• High pressure pneumatic operation• Screw press for fine pressure adjustment• High-quality needle valve for metering in supply
pressure

P5514 Hydraulic Comparison Pump
• Pressure range: 0 to 70 MPa (10 000 psi)
• Bench mount design• Screw pressure for pressure generation• Operates with a wide range of fluids (Viton seals
standard, EP seals available)

P5515 Hydraulic Comparison Pump
• Pressure range: 0 to 140 MPa (20 000 psi)
• Built-in priming pump for large volume applications
and system priming• High-quality screw press for pressure generationand fine control• Clear acrylic reservoir for visibility of fluid level andquality• Built-in drain plug for easy removal of fluid• Sturdy carrying case with lid• Operational with a wide range of fluids (Viton seals
standard, EP seals available)

2 Fluke Corporation

2700G Series Reference Pressure Gauges

ModelRange

Resolution

2700G-BG100K-15 psi to 15 psi-100 kPa to 100 kPa

0.001 psi
0.0001 bar

Uncertainty (k=2) Burst pressure
45 psi0.3 MPa

Note 12700G-BG200K-15 to 30 psi-100 kPa to 200 kPa

0.001 psi
0.0001 bar

90 psi0.6 MPa

Note 12700G-BG700K-12 psi to 100 psi-80 kPa to 700 kPa

0.01 psi
0.0001 bar

1000 psi
7 MPa

Note 22700G-BG2M-12 psi to 300 psi-80 kPa to 2 MPa

0.01 psi
0.001 bar2000 psi
14 MPaNote 22700G-BG3.5M-12 psi to 500 psi-80 kPa to 3.5 MPa

0.01 psi
0.001 bar2000 psi
14 MPaNote 22700G-BG7M-12 to 1000 psi-80 kPa to 7 MPa

0.1 psi0.001 bar

10 000 psi
70 MPaNote 22700G-G20M

0 to 3000 psi
0 to 20 MPa

0.1 psi0.01 bar

10 000 psi
70 MPaNote 32700G-G35M

0 to 5000 psi
0 to 35 MPa

0.1 psi0.01 bar

10 000 psi
70 MPaNote 32700G-G70M

0 to 10 000 psi
0 to 70 MPa

1 psi
0.01 bar

15 000 psi
100 MPa

Note 3

Positive pressure± 0.02 % FSVacuum± 0.05 % FS

Media compatibility

Uncertainty includes linearity, hysteresis, repeatability, resolution, reference uncertainty, 1 year long term stability, andtemperature effects from 18 to 28 °C.
For temperatures from 0 °C to 18 °C and 28 °C to 50 °C, add 0.003% FS/°CMedia compatibility:
Note 1: Any clean, dry non-corrosive gas
Note 2: Any liquids or gases compatible with 316 stainless steel
Note 3: Any non-flammable, non-toxic, non-explosive, non-oxidizing liquid or gas compatible with 316 stainless steel

Mechanical specifications
Dimensions (HxWxD)

12.7 cm x 11.4 cm x 3.7 cm (5 in x 4.5 in x 1.5 in)

Pressure connection¼ in NPT Male

Housing/weight

Cast ZNAL/0.56 kg (1.22 lb)

Display

5-1/2 digits, 16.53 mm (0.65 in) high
20-segment bar graph, 0 to 100 %Power

Three AA alkaline batteries

Battery life

75 hours

3 Fluke Corporation

2700G Series Reference Pressure GaugesOrder information
Model

Description

2700G-BG200K

Reference Pressure Gauge, -15 to 30 psi(-100 to 200 kPa)

2700G-BG100K

2700G-BG700K
2700G-BG2M
2700G-BG3.5M
2700G-BG7M
2700G-G20M
2700G-G35M
2700G-G70M
P5510-2M
P5513-20M
P5514-70M
P5514-70M-EP
P5515-140M
P5515-140M-EP

Fluke Calibration.

Precision, performance, conﬁdence.™

Reference Pressure Gauge, -15 to 15 psi(-100 to 100 kPa)

Reference Pressure Gauge, -12 to 100 psi(-80 to 700 kPa)
Reference Pressure Gauge, -12 to 300 psi(-80 kPa to 2 MPa)
Reference Pressure Gauge, -12 to 500 psi(-80 kPa to 3.5 MPa)

Reference Pressure Gauge, -12 to 1000 psi(-80 kPa to 7 MPa)
Reference Pressure Gauge, 0 to 3000 psi(0 to 20 MPa)
Reference Pressure Gauge, 0 to 5000 psi(0 to 35 MPa)

Fluke Calibration
PO Box 9090, Everett, WA 98206 U.S.A.

Reference Pressure Gauge, 0 to 10 000 psi(0 to 70 MPa)

Fluke Europe B.V.
PO Box 1186, 5602 BD
Eindhoven, The Netherlands

Pneumatic Comparison Test Pump, vacuum to300 psi (2 MPa)

For more information call:In the U.S.A. (877) 355-3225 orFax (425) 446-5116
In Europe/M-East/Africa +31 (0) 40 2675 200 or
Fax +31 (0) 40 2675 222
In Canada (800)-36-FLUKE or
Fax (905) 890-6866
From other countries +1 (425) 446-5500 or
Fax +1 (425) 446-5116
Web access: http://www.ﬂukecal.com

Pneumatic Comparator, 0 to 3000 psi (20 MPa)
Hydraulic Comparison Pump with Viton Seals,
0 to 10 000 psi (70 MPa)
Hydraulic Comparison Pump with EP Seals,
0 to 10 000 psi (70 MPa)

Hydraulic Comparison Pump with Viton Seals,
0 to 20 000 psi (140 MPa)©2012-2013 Fluke Calibration.
Speciﬁcations subject to change without notice.
Printed in U.S.A. 3/2013 4264507A̲ENPub-ID: 12005-eng Rev 01

Hydraulic Comparison Pump with EP Seals,
0 to 20 000 psi (140 MPa)

Modiﬁcation of this document is not permitted
without written permission from Fluke Calibration.

Items included with gauge:• Manual on CD-ROM in 14 languages• Calibration certiﬁcate showing traceability• 1/4 NPT female to 1/4 BSP male adapter• 1/4 NPT female to M20X1.5 adapter• Universal ac power supply• USB communication cable

Optional:

ISO 17025 accredited calibration available(add “/C” to the model number)

4 Fluke Corporation

2700G Series Reference Pressure Gauges

株式会社大手技研
本社 ：茨城県つくば市観音台1-25-12
TEL : 029-839-0777

FAX : 029-839-2288

テクノロジーセンター ：茨城県つくば市観音台1-25-12
TEL : 029-839-0778

FAX : 029-839-4488

関 西 営 業 所 ：兵庫県明石市松の内2-1-8 50ヤングビル6F
TEL : 078-926-1178

FAX : 078-926-1180

ホームページ

https://www.ohtegiken.co.jp

E-Mailmain.sales@ohtegiken.co.jp