Page 28

AXIAL FORCE AND TORQUE(SWT) SERIES TRANSDUCERSAXIAL FORCE AND TORQUE (SWT)SERIES TRANSDUCERSFEATURES• Measure both Force and Torque• High Accuracy Both Bridges• Low Crosstalk• Low Creep• 5pt T/C Axial & CW/CCW Torsion Calibration• Proprietary Modulus Compensated Strain Gages

SPECIFICATIONSSPECIFICATIONS
ModelCapacity

U.S. lbf./ in-lb

PARAMETERSDIMENSIONS
Model

300/150, 500/250,
1K/500,inch

SWT-10

SWT-202K/1K, 5K/2K
inch

10K/3K
inch

1.

3.63

3.63

4.7

2.

R 2.53R 2.53R 3.5

3.

4.12 Dia.

4.12 Dia.

6.06 Dia

Capacity
Dimensions

SWT-10

SWT-10

SWT-20300/150, 500/250, 1K/500,
Axial Force
BridgeTorque Bridge

2K/1K, 5K/2K
Axial Force
BridgeTorque Bridge

10K/3K
Axial Force
BridgeTorque Bridge+/-0.040+/-0.070+/-0.040+/-0.070+/-0.040+/-0.070+/-0.040+/-0.050+/-0.040+/-0.050+/-0.040+/-0.050+/-0.020+/-0.050+/-0.020+/-0.050+/-0.020+/-0.050+/-0.025+/-0.025+/-0.025+/-0.025+/-0.025+/-0.025+/-2.0+/-2.0+/-2.0+/-2.0+/-2.0+/-2.0+15 to +115

+15 to +115

+15 to +115

+15 to +115

+15 to +115

+15 to +115

-10 to +45-10 to +45-10 to +45-10 to +45-10 to +45-10 to +45-65 to +200

-65 to +200

-65 to +200

-65 to +200

-65 to +200

-65 to +200

-55 to +90-55 to +90-55 to +90-55 to +90-55 to +90-55 to +90

SWT-104.

3/8 -24
3/8 -24
3/8-24.66 Deep on 3.00 B.C. .66 Deep on 3.00 B.C. .66 Deep on 3.00 B.C.

5.

.25 SF on 3.00 B.C.

.25 SF on 3.00 B.C.

.25 SF on 3.00 B.C.

6.

5/8-18.95 Deep

5/8-18.95 Deep

5/8-181.05 Deep

ACCURACY

Nonlinearity, % Rated Output
Hysteresis, % Rated Output
Nonrepeatability, % Rated Output
Creep, % in 20 minutes
Zero Balance, % Rated Output

3TEMPERATURE
Range, Compensated, °F
Range, Compensated, °C
Range, Operating, °F
Range, Operating, °C
Effect on Sensitivity, % Rdg/ °F
Effect on Sensitivity, % Rdg/ °C
Effect on Zero, % Rated Output/ °F
Effect on Zero, % Rated Output/ °C+/-0.0008+/-0.0008+/-0.0008+/-0.0008+/-0.0008+/-0.0008+/-0.0015+/-0.0015+/-0.0015+/-0.0015+/-0.0015+/-0.0015+/-0.0015+/-0.0015+/-0.0015+/-0.0015+/-0.0015+/-0.0015+/-0.0027+/-0.0027+/-0.0027+/-0.0027+/-0.0027+/-0.0027

1.0

1.5

2.0

2.5

2.0

1.0

20

20

20

20

20

20

350

350

350

350

350

350

350

350

350

350

350

350

5000

5000

5000
5000
PT02E-10-6P(MS3112E-10-6P)

5000

5000

7

OPTIONS
4

2

Bayonet ConnectorMECHANICAL
Safe Overload Range, % Capacity
Weight lb
Flexure Material

Integral cable
Custom connectors

1Connector Protectors

ACCESSORIES
564Mating connector
Loading fixtures
Interconnect cable
Shunt calibration resistor+/-150+/-150+/-150+/-150+/-150+/-150

4.15

4.15

10.6

10.6

32.19

32.19

Aluminum

Output adjustment to standard
Compression overload protection

5ELECTRICAL
Rated Output, mV/V, Nominal
Excitation Voltage, VDC Maximum
Input Resistance, Ohms
Output Resistance, Ohms
Insulation Resistance, Megohm

Additional bridges

Instrumentation

SteelTOVEY ENGINEERING, INC. • 1619 W Knudsen Drive • Phoenix, AZ 85027
Phone: (623) 434-5110 • Fax: (623) 434-5130
Email: tei_sales@toveyengineering.com • Web: www.toveyengineering.com© 2015 Tovey Engineering, Inc.

TOVEY ENGINEERING, INC. • 1619 W Knudsen Drive • Phoenix, AZ 85027
Phone: (623) 434-5110 • Fax: (623) 434-5130
Email: tei_sales@toveyengineering.com • Web: www.toveyengineering.com© 2015 Tovey Engineering, Inc.

Page 29

Page 28

AXIAL FORCE AND TORQUE(SWT) SERIES TRANSDUCERSAXIAL FORCE AND TORQUE (SWT)SERIES TRANSDUCERSFEATURES• Measure both Force and Torque• High Accuracy Both Bridges• Low Crosstalk• Low Creep• 5pt T/C Axial & CW/CCW Torsion Calibration• Proprietary Modulus Compensated Strain Gages

SPECIFICATIONSSPECIFICATIONS
ModelCapacity

U.S. lbf./ in-lb

PARAMETERSDIMENSIONS
Model

300/150, 500/250,
1K/500,inch

SWT-10

SWT-202K/1K, 5K/2K
inch

10K/3K
inch

1.

3.63

3.63

4.7

2.

R 2.53R 2.53R 3.5

3.

4.12 Dia.

4.12 Dia.

6.06 Dia

Capacity
Dimensions

SWT-10

SWT-10

SWT-20300/150, 500/250, 1K/500,
Axial Force
BridgeTorque Bridge

2K/1K, 5K/2K
Axial Force
BridgeTorque Bridge

10K/3K
Axial Force
BridgeTorque Bridge+/-0.040+/-0.070+/-0.040+/-0.070+/-0.040+/-0.070+/-0.040+/-0.050+/-0.040+/-0.050+/-0.040+/-0.050+/-0.020+/-0.050+/-0.020+/-0.050+/-0.020+/-0.050+/-0.025+/-0.025+/-0.025+/-0.025+/-0.025+/-0.025+/-2.0+/-2.0+/-2.0+/-2.0+/-2.0+/-2.0+15 to +115+15 to +115+15 to +115

+15 to +115

+15 to +115

+15 to +115

-10 to +45-10 to +45-10 to +45-10 to +45-10 to +45-10 to +45-65 to +200

-65 to +200

-65 to +200

-65 to +200

-65 to +200

-65 to +200

-55 to +90-55 to +90-55 to +90-55 to +90-55 to +90-55 to +90

SWT-104.

3/8 -24
3/8 -24
3/8-24.66 Deep on 3.00 B.C. .66 Deep on 3.00 B.C. .66 Deep on 3.00 B.C.

5.

.25 SF on 3.00 B.C.

.25 SF on 3.00 B.C.

.25 SF on 3.00 B.C.

6.

5/8-18.95 Deep

5/8-18.95 Deep

5/8-181.05 Deep

ACCURACY

Nonlinearity, % Rated Output
Hysteresis, % Rated Output
Nonrepeatability, % Rated Output
Creep, % in 20 minutes
Zero Balance, % Rated Output

3TEMPERATURE
Range, Compensated, °F
Range, Compensated, °C
Range, Operating, °F
Range, Operating, °C
Effect on Sensitivity, % Rdg/ °F
Effect on Sensitivity, % Rdg/ °C
Effect on Zero, % Rated Output/ °F
Effect on Zero, % Rated Output/ °C+/-0.0008+/-0.0008+/-0.0008+/-0.0008+/-0.0008+/-0.0008+/-0.0015+/-0.0015+/-0.0015+/-0.0015+/-0.0015+/-0.0015+/-0.0015+/-0.0015+/-0.0015+/-0.0015+/-0.0015+/-0.0015+/-0.0027+/-0.0027+/-0.0027+/-0.0027+/-0.0027+/-0.0027

1.0

1.5

2.0

2.5

2.0

1.0

20

20

20

20

20

20

350

350

350

350

350

350

350

350

350

350

350

350

5000

5000

5000
5000
PT02E-10-6P(MS3112E-10-6P)

5000

5000

7

OPTIONS
4

2

Bayonet Connector
MECHANICAL
Safe Overload Range, % Capacity
Weight lb
Flexure Material

Integral cable
Custom connectors

1Connector Protectors

ACCESSORIES
564Mating connector
Loading fixtures
Interconnect cable
Shunt calibration resistor+/-150+/-150+/-150+/-150+/-150+/-150

4.15

4.15

10.6

10.6

32.19

32.19

Aluminum

Output adjustment to standard
Compression overload protection

5ELECTRICAL
Rated Output, mV/V, Nominal
Excitation Voltage, VDC Maximum
Input Resistance, Ohms
Output Resistance, Ohms
Insulation Resistance, Megohm

Additional bridges

Instrumentation

Steel

TOVEY ENGINEERING, INC. • 1619 W Knudsen Drive • Phoenix, AZ 85027
Phone: (623) 434-5110 • Fax: (623) 434-5130
Email: tei_sales@toveyengineering.com • Web: www.toveyengineering.com© 2015 Tovey Engineering, Inc.

TOVEY ENGINEERING, INC. • 1619 W Knudsen Drive • Phoenix, AZ 85027
Phone: (623) 434-5110 • Fax: (623) 434-5130
Email: tei_sales@toveyengineering.com • Web: www.toveyengineering.com© 2015 Tovey Engineering, Inc.

Page 29