Bioquell ProteQ®

Mobile bio-decontamination
system for rooms and zones

GLOB AL E X PER T S I N
C ON TAM I N ATION CO N T R O L

BioquellProteQ®

An advanced mobile
bio-decontamination systemwith faster cycles andgreater ease of use
From research laboratories to biopharmaceutical manufacturers,
routine or emergency remediation is made possible with automatedbio-decontamination technology from Ecolab. The Bioquell ProteQ systemprovides a mobile and innovative delivery system utilising Ecolab’s Bioquell
35% Hydrogen Peroxide Vapour bio-decontamination technology.

MOBILE, VERSATILE
AND WIRELESSWith all components stored within
its mobile frame, quickly dispatch thesystem in one trip by a single person.
Wireless communication between
components makes setup easy

BPR APPROVED
Bioquell HPV-AQ 35% hydrogen
peroxide holds a European Union
authorization under the Biocidal
Products Regulation (BPR,Regulation (EU) 528/2012)

The Bioquell ProteQ delivers powerful
distribution to help reduce cycle
times and cover areas as large as
400m3 for each system used. It also
communicates wirelessly to thecontrol lectern positioned outside ofthe area being bio-decontaminated
so that cycles can be started andmonitored remotely.
The Bioquell ProteQ is a compact
and mobile system that offers builtin aeration, but cycle times can beoptimised further with the addition of
optional wirelessly controlled aeration
units which can be stored directlyin the system’s frame for added
convenience.
Applications:manufacturing areas◥ GMP/GLP Laboratories◥ Biosafety labs of all levels◥ Animal facilities

No additional wiping or cleaning
required after each cycle◥ Automated cycles◥ 6-log sporicidal kill on everyexposed surface◥ RFID enabled bottles fortraceability and quality control◥ 21 CFR Part 11 audit trail

compliance package available◥ Optional aeration units for evenquicker bio-decontamination cycles◥ Single system covers 400m3. Up tothree systems can be connected
for larger areas◥ Wireless networking for rapid setup◥ Paper and digital cycle reports tosupport GMP compliance◥ Biopharmaceutical

AUTOMATED CYCLES

Bioquell ProteQ offers:◥ Production areas withdifficult to clean spaces
such as RABS
◥ Cleanrooms, lab spacesand anterooms◥ Other room and zoneneedsComponents

11The main structure of thesystem contains:
The hydrogen peroxide
dispersion and distribution
system for rooms up to 400 m³(depending on configuration, loadand environmental conditions).
Dual bottle module with RFID
reading for safe loading, large
volume decontamination and
key data collection.

3

Built-in aeration, capability
comes as a standard feature in
every Bioquell ProteQ.

22

The wireless control module
combines a thermal printerand a colour touch screen withgraphical interface.

3

Additional aeration units
help increase airflow during
the bio-decontamination cycle
and help reduce cycle times byremoving hydrogen peroxide
vapour during the aerationphase. These units are stored
within the frame of the main
structure for ease of transportand setup.

Large space Bio-decontamination
made easy
When the need arises forbio-decontaminating areas
greater than 400m3, users cannetwork up to three systems
together quickly and easily using
the multi-system controller option,
making the Bioquell ProteQ ideal
for nearly any size area.

Your Bioquell ProteQ system can be optimized with:◥ Process monitoring sensor package◥ System commissioning and IQ/OQ/PQ◥ 21 CFR Part 11 audit trail◥ Master control panel for simultaneous operation of up

compliance package◥ Low level hydrogen peroxide

safety sensor◥ Laser room dimension measuring device◥ Additional wireless aeration units

to three systems◥ Integration to Building Management System (BMS)◥ Advanced validation such as gas cycle development(GCD) and Performance Qualification (PQ)
◥ Preventative Maintenance plans

Bioquell® ProteQ
Technical Specifications
ProteQ

Physical and safety data

Operating data

Power data

Dimensions and weight:
with lectern 672 x 1430 x 710 mm(Height = 1421 mm if no lectern)

Operating Temperature:
Operate at ambient temperatures1

Regional power requirements (max):
230 V single phase, 50/60Hz 7.5A
Base Unit, 1A lectern
120 V single phase, 50/60Hz 15A
Base Unit, 2A lectern
100 V single phase, 50/60Hz 15A
Base Unit, 2A lecternLectern:
630 x 880 x 349 mm

Relative Humidity limits at startof cycle: 10% to 80% RH (Max)2
Enclosure decontamination capacity:
Nominally 400 m3 for rooms (subject toconfiguration, loading andenvironmental conditions)

Mass:with lectern: 67 Kg
Lectern: 9 Kg

Maximum H202 liquid volume: 4 litres
2 positions for 2 litre or 950ml bottles

Max power consumption(Main Unit/Lectern) 1.8 kW / 0.2kW
Power Supply:
Installation category II

For concerns or questions related to temperature limits, please contact Ecolab.

1	2

If requiring to operate outside this range refer to Ecolab or its agents.

For regional oﬀice details, please visit bioquell.com
ECOLAB LTD
52 Royce Cl
Andover
SP10 3TS, UKEUROPE HEADQUARTERS
Richtistr. 78304 Wallisellen
Switzerland

www.bioquell.com

www.ecolab.com/lifesciences© 2023 Ecolab USA Inc. All rights reserved. 31MAY23/EUbioquell code TBD11604.2

USE BIOQUELL PRODUCTS SAFELY. ALWAYS READ THE LABEL AND PRODUCT INFORMATION BEFORE USE.