7000/7800/9000

REUSABLE RESPIRATORSFITTING INSTRUCTIONS AND
SELECTION GUIDE
7000/7800 HALF MASKSSTANDARD LOCK DOWN MODE

9000 FULL-FACE MASKCARTRIDGES & FILTERSDROP DOWN MODE

7100
Organic Vapor
Cartridge
7200
Acid Gas Cartridge
Chlorine/Hydrogen Chloride/Sulfur
Dioxide/or Chlorine Dioxide/or Hydrogen
Flouride/or Hydrogen Sulfide.

To lock the straps in position, open the valve
cover by pulling up on the latch. Position
straps underneath the tabs of the strap
channel. Close and snap shut the valve cover.

Open valve cover by pulling up on the latch.
Pull the straps out from underneath the tabsof the strap channel. Close and snap shutthe valve cover.

Grab lower lengths of head harness witheach hand, just above each adjustment
buckle, and while positioning chin in flange,pull harness over the head.

7300
Organic Vapor/
Acid Gas Cartridge
Organic Vapors/Chlorine/Hydrogen
Chloride/Sulfur Dioxide/Or Chlorine
Dioxide/Or Hydrogen Fluoride/
Or Hydrogen Sulfide.

7400
Ammonia/
Methylamine
Cartridge
First adjust head cradle then place respirator
under the chin and pull the cradle to the topof the head so it rests on the crown.

Attach the bottom straps directly against
the back of your neck.

Push facepiece onto face and chin with one
hand and position for most comfortable andsecure seal while pulling and tightening
neck tab of harness with other hand. Repeat
for other neck tab. For easiest, most secureand best fit, always adjust neck harness
tabs first before tightening top harness tabs.

7500
Formaldehyde
Cartridge
7600
Multi-Gas/Vapor Smart®
Cartridge

PERFORM BOTH NEGATIVE
AND POSITIVE SEAL CHECKSEACH TIME BEFORE ENTERINGA CONTAMINATED AREA.
Attach the bottom straps directly against
the back of the neck.

Organic Vapors/Ammonia/Methylamine/
Chlorine/Hydrogen Chloride/SulfurDioxide/Or Chlorine Dioxide/
Or Hydrogen Fluoride/
Or Formaldehyde/Or Hydrogen Sulfide.

Adjust head cradle then pull the cradle tothe top of the head so it rests on the crown.

7990
P100 Filter Cartridge

Tighten top straps first then the lower neckstraps until you have a very snug andleak-free fit. Facepiece should not be loose
or slide up or down the nose. Tighten bypulling on ends or loosen by pushing out onbuckle tab. Do not overtighten. Further
tighten top straps if necessary.

Tighten top straps first then the lower neckstraps until you have a very snug andleak-free fit. Facepiece should not be loose
or slide up or down the nose. Tighten bypulling on ends or loosen by pushing out onbuckle tab. Do not overtighten. Further
tighten top straps if necessary.

Negative Pressure Seal Check: Coverboth cartridges without pressing too hardagainst face, then gently inhale and hold yourbreath. The facepiece should slightly
collapse. If air leakage is detected, re-adjust
the position of the facepiece and the tensionof head harness and repeat the seal check
until leakage is eliminated.

7940
P100 Filter Disk

7950
P100 plus Nuisance
OV/AG Filter Disk

7960
P100 plus Nuisance
OV Filter DiskPERFORM BOTH A NEGATIVE AND POSITIVE SEAL CHECKEACH TIME BEFORE ENTERING A CONTAMINATED AREA.

Negative Pressure Seal Check: Coverboth cartridges without pressing too hardagainst face, and gently inhale and holdyour breath. The facepiece should slightlycollapse. If air leakage is detected, re-adjust
the position of the facepiece and the tensionof both headstraps and repeat the seal
check until leakage is eliminated.

Positive Pressure Seal Check: Coverthe exhalation valve vent without pressing
too hard against face, and exhale gently tocreate a slight positive pressure. If airleakage is detected, re-adjust the positionof the facepiece and the tension of bothheadstraps and repeat the seal check until
leakage is eliminated.

Positive Pressure Seal Check: Cover theexhalation valve vent without pressing too
hard against face, then exhale gently tocreate a slight positive pressure. If airleakage is detected, re-adjust the positionof the facepiece and the tension of headharness and repeat the seal check until
leakage is eliminated.

8910 N95 Pre-Filter

8970 R95 Pre-Filter

7140 Organic Vapor
Cartridge/
P100 Cartridge Disk

7640 Multi-Gas/Vapor
Smart® Cartridge/
P100 Cartridge Disk

7000/7800/9000 FACEPIECE, ACCESSORIES AND REPLACEMENT PARTS
7020
Pre-Filter Retainer

7920
Piggyback Adapter

7999
Splash/Spark Protector

7001/7801/9001 Facepiece, Small
7002/7802/9002 Facepiece, Medium
7003/7803/9003 Facepiece, Large
WARNING: Before using these respirators, you must read manual for complete instructions. If you cannot obtain a proper fit, do not enter the contaminated area and see your supervisor.
Follow all instruction and warnings on the use of this respirator and wear at all times of exposure. Failure to do so will reduce respirator effectiveness, wearer protection, and may result insickness or death. For additional information, consult your Supervisor, read all information on box and phone +1 (800) 421-0668 or +1 (310) 837-6500 ext. 512/550. www.moldex.com
9900-046 REV G 5/20

9000 Accessories andReplacement Parts:
0073 Diaphragm Kit
0074 Gasket Kit
0092 Head Harness
0093 Face Shield Protectors
0094 Med./Large Nose Cup Assembly
0095 Small Nose Cup Assembly
0098 Spectacle Kit
9780 Reusable Respirator Bag

7000 Replacement Parts:
0072 Head Harness Assembly
0073 Diaphragm Kit
0074 Gasket Kit