Waterproof Platform Scales

FG-CWPSeries

M-size platform (30 kg)

L-size platform (60 kg / 150 kg)

Weigh efficiently,and wash eﬀectively

www.aandd.jp

Introducing A&D’s Highly Washable
All Stainless Steel Platform Scales
While realizing fast and accurate weighing, A&D’s FG-CWP series of new
waterproof platform scales allow cleaning with alcohol as well as rinse withwater. This makes the series suitable for use in environments where avoiding
hazards of contamination is imperative, such as food-processing factories.

IP67 dust & waterproof
In accordance with the IP67 rating of the IEC/EN standard 60529, the FG-CWP series is designed
to prevent the intrusion of water even while immersed to a depth of 1 meter for 30 minutes. Thescale also permits no particulate matters such as wheat ﬂour or seasoned powder, which candamage the device, inside the housing.

“Also being easy to wash is critical for hygiene control.”All stainless-steel construction
The stainless steel*1 surface is resistant to rust/corrosion and scratches. Also being easy to wash
is critical for hygiene control. Further, the display unit (including the front panel), weighing pan,pole and frame can be cleaned using alcohol.*2

High-speed measurement of 1 second*3unit, weighing pan, pole and frame: SUS304*1 Display
Load cell: SUS420J-2 with protective coating*2 Diluted to 70% or less

The fast, eﬃcient weighing increases productivity while signiﬁcantly reducing operator frustration.
when the readability is set to normal*3 Typical(factory setting)

P.1

“You can easily save time and costs with these lights.”
Bright and multi-colored LED comparator lights
With highly intuitive comparator lights, checkweighing (judging whether the measured weight iswithin the designated limits) becomes far moreaccurate and eﬃcient than with a numeric display
alone. For later use and simple changeover, the scale
stores up to 10 comparator limit settings. You caneasily save time and costs with these lights.

•••••■ Auto-tare and negative comparison functions
The scale can perform tare automatically when the weight reaches the set target range.
Additionally, negative values can be used as comparator limits. With these functions combined,
you can perform "take-away" checkweighing, which is a weighing method where a target amountof a substance is repeatedly removed from the container with no key press.

USB cable (3 m) for both power supply and data communication
The FG-CWP series receives power supply via USB (Type-A plug), which can be connected toeither an AC adapter (provided as standard), a USB port of a PC, or an oﬀ-the-shelf mobile
battery.*4 Further, when connected to a PC, bi-directional data communication between the scale
and the PC is possible using Virtual COM.*5 (The WinCT data communication software isavailable for free download from A&D's website.)reference, continuous operation time with an oﬀ-the-shelf mobile battery*4 For(10,000 mAh) is estimated to be approx. 77 hours with backlight on.
*5 Installation of the dedicated driver software is required for Windows 8.1 or earlier.

Other useful features✔ Large LCD digits (30 mm character height) with white backlight that provides excellent
visibility in dim light✔ Four selectable weighing resolutions: 1/3,000, 1/6,000 (or 1/7,500), 1/12,000 (or 15,000), and 1/30,000✔ Display hold function that makes it easy to read and record the display value when external
disturbances can occur unexpectedly✔ Adjustable response characteristics to optimize performance under the given environmental
condition (e.g. vibration) at the location✔ Counting mode to count a number of small pieces of thesame weight instantly and precisely✔ Freely adjustable display angle to suit the convenience ofindividual operators✔ Password lock and key restriction functions to prevent
operators from making unwanted changes to the scalesettings✔ Impact shock detection (ISD) function that helps avoid
errors or damage to the load cell due to impact loads
Freely adjustable display angle
P.2

Speciﬁcations
kgglbozlb-ozCounting mode

FG-30KCMWP

FG-60KCLWPCapacity

30

60

150

Readability

0.001 / 0.002 / 0.005*i / 0.01

0.002 / 0.005 / 0.01*i / 0.02

0.005 / 0.01 / 0.02*i / 0.05

Capacity

30000

Readability

1 / 2 / 5*i / 10

FG-150KCLWP

N/ACapacity

66

130

330

Readability

0.002 / 0.005 / 0.01*i / 0.02

0.005 / 0.01 / 0.02*i / 0.05

0.01 / 0.02 / 0.05*i / 0.1Capacity

1050

2100

5200

Readability

0.05 / 0.1 / 0.2*i / 0.5

0.1 / 0.2 / 0.5*i / 1

0.2 / 0.5 / 1*i / 2

Capacity

66

130

330

Readability

0.1

11Max. countable number

300000

Number of samples

5, 10, 20, 50 or 100

Repeatability (std. deviation) [kg]

0.005

0.01

0.02

Linearity [kg]

±0.01

±0.02

±0.05

Stabilization time

Approx. 1 sec.*ii

Sensitivity drift±20 ppm / ˚C (5 to 35 ˚C / 41 to 95 ˚F)
7-segment LCD (character height: 30 mm) with backlight

Display

3-color, 3-step LED comparator lights

Display refresh rate

Approx. 10 times / second

Operating environment–10 ˚C to 40 ˚C / 14 ˚F to 104 ˚F, 85% R.H. or less (no condensation)
AC adapter (provided as standard), USB port*iii or mobile battery*iv via USB Type-A (5 V / 100 mA)

Power supply

USB cable length: Approx. 3 m

Weighing pan size
Dimensions (W × D × H)

300 × 380 mm

370 × 500 mm

11.81 × 14.96 in

14.57 × 19.69 in

300 × 515 × 784 mm

370 × 635 × 784 mm

11.81 × 20.28 × 30.87 in

14.57 × 25 × 30.87 in11.1 kg / 24.5 lb

12.8 kg / 28.3 lb

Weight (approx.)

setting*iii Factorywhen the readability is set to normal (factory setting)*iii Typicalcommunication is also possible when connected to a USB port of a PC.
*iv Data
reference, continuous operation time with an oﬀ-the-shelf mobile battery (10,000 mAh) is estimated to be approx. 77 hours with backlight on.
* For(Operation is not guaranteed for all mobile batteries.)

Options
FG-27CWP

Bluetooth® communication interface for FG-CWP*v*v For bi-directional data communication with a PC, the AD-8541-PC dongle (sold separately) is required.

IP67

LCD
Backlight

Bi-directional
USB

User AccessControl

GLP/GMP
CompliantImpact ShockDetection

Bluetooth®
Interface*vi

Rechargeable
Battery*vii

CountingFunction

Optional
Comparator
Function

AutoPower ONAuto
Power OFF

Bluetooth interface can be currently*vi Theused in the US, Canada and Japan only.
®

*vii Commercially-available mobile battery

A&D Company, Ltd. (JAPAN)

A&D Korea Ltd. (South Korea)

A&D Engineering, Inc. (USA)A&D Rus Co., Ltd. (Russia)

A&D Australasia Pty Ltd. (Australia)

A&D Instruments India (P) Ltd. (India)A&D Instruments Ltd. (United Kingdom)

A&D Scientech Taiwan Ltd. (Taiwan)<German Sales Office>A&D Instruments Thailand Ltd. (Thailand)

URL: aandd.jp

URL: andonline.com

URL: andaustralasia.com.au
URL: andprecision.com
URL: andprecision.com

URL: andk.co.kr
URL: and-rus.ru

URL: aanddindia.in

URL: aandd.com.tw

URL: thai.andprecision.com* FG-CWP-ADCC-01-PR1-225