TRISON 4000.2

Ultrasonic bath for robotic, MISand standard instruments

A new standard for the pre-cleaning

1Complex instruments = complex
reprocessing?
Due to their complex structure, robotic
instruments have many critical points
that need to be reprocessed. The instruments, which cannot be disassembled, consist of a housing with a long,thin shaft, various filigree Bowden cables and complex tips.
Due to their design and surgical use,
robotic instruments need to be reprocessed via many individual process
steps in order to be used on the next
patient. Here, particular attention ispaid to effective pre-cleaning before
the actual disinfection and sterilisation
processes. Purely manual pre-cleaning,which not only ties up a lot of staff due
to the repeated manual rinsing, moving and cleaning but is also prone toerrors, is not acceptable for users as areprocessing step.

Before reprocessing.

After reprocessing with ultrasound.

The alternative: an ultrasonic bath,specially developed for pre-cleaning robotic instruments – the TRISON 4000.2.

Robot-assisted surgery?
That‘s no longer a vision of the future!
The robotic systems now used in many
clinics for supporting surgical treatments offer surgeons numerous advantages. First of all, they ensure a gentle
and highly precise, minimally invasive
procedure that can be carried out while
preserving the nerves and organs.

2

For patients, procedures are more tolerable and put less strain on the body,and the recovery time is shortened. The
highly complex technology has long become a standard in modern surgery andwill come to be even more relevant forother medical disciplines in the future.

3

TRISON 4000.2: the standard forpre-cleaning robotic instruments
It is an innovative, modular ultrasonic bath, adapted torobotic instruments, which makes it possible to performthe 3 basic functions of cleaning, rinsing and moving in
one process. This sequence ensures the best, and reproducible, cleaning results, saves time during pre-cleaning,
optimises the overall reprocessing procedure, and also
provides valid process documentation. Intuitive operation
via the large touch display also ensures safe handling.

Movement
Improves cleaning performance

The patented moving device is designed
to hold 4 instruments at the same time andallows each individual instrument to movethroughout the entire process. Incrustations
can thereby be loosened, and all accessible
points of the instruments are reached byultrasound.

UltrasoundAchieves the best possible cleaning
results
The ultrasonic oscillating tank, whose
dimensions are designed for large, robotic
instruments, enables time-saving andintensive, but also gentle, instrument
cleaning and is particularly effective in
hard-to-reach places.

For more information and videos:
youtube.com/@BANDELIN/videos
4Rinsing
Ensures effective interior cleaning

The instruments are rinsed several timesduring the process in several sequential procedures. Contamination is removed from theinstrument, and, using single-channel control,
the rinsing control checks to see whether theinstruments are rinsable.

5Easy to use – safe processes
Everything at a glance
The programs can be selected in one click directlyon the home screen. Other useful functions, suchas rapid degassing, device care and settings, canbe easily selected thanks to large buttons withsupporting icons.

Operation
via
7-inch touch
display

User-optimised menu navigation:• Program quick start or supporting program management• Visual representation of individual steps• Help pages with explanations• Individual settings (language, process times, warning temperatures, etc.)• Program favourites• Initial configuration when putting into service• Access for servicing

Process screen with successful rinsing.

Visual representation of individual steps.

Process documentation
The TRISON 4000.2 has an interface concept for
digital process documentation. A protocol with all
relevant cleaning parameters is created for eachindividual cleaning process. The protocols can betransmitted via the USB interface. Alternatively, theTRISON can be integrated into CSSD management
software via the integrated network interface.

.TXT67

1 device – 3 uses
The TRISON has been specially developed for robotic instruments. However,MIS and standard instruments can also
be cleaned effectively with appropriate accessories. In order to meet theOther advantages:

cleaning requirements in question in
the best possible way, there is a cleaning program available for each type ofinstrument.

Temperature monitoringwith warning function
Ensures that the set limit
values are complied with.

User languages
Easy to select different
language outputs.

Flexible mounting
Available as a left-sided orright-sided version – for integration into a row or as a single
solution in a rollable cabinet.

Robotics

MIS

Standard

For simultaneous cleaning of up to
4 robotic instruments using the Twist
moving device and corresponding
hose connections for rinsing.

Ultrasonic cleaning and sequential individual rinsing of up to 8 MIS instrumentsin the specially developed TRISON Rack.
The MIS program also includes rinsing
channel monitoring.

For cleaning instruments in DIN andISO baskets before further machine reprocessing. A basket holder is required.

Process safety due toself-monitoring
Control of all functions toavoid incorrect operation.

Overall process optimisation
Increases the quality andreproducibility of the entire
treatment chain.

89Integrated or mobile?
The TRISON 4000.2 is available in two versions: as abuilt-in device or as a mobile device in a rollable cabinet,
the SONOBOARD TRISON.
For optimal workflows, the best possible variant is to integrate the TRISON
into a stainless steel row of units. Depending on the orientation of the precleaning area, the ultrasonic bath canbe installed on the left or right side.
However, the space conditions inCSSDs are often limited, or it is notpossible to convert the stainless steel
row at the particular time in question.
In such cases, the SONOBOARD variantof the TRISON can be procured. The

Fits into
every
pre-cleaningarea

SONOBOARD is a ready-to-use set,i.e., all components are preassembled
in a rollable stainless steel cabinet.
When putting the device into service,
only the necessary media (electricity,waster water system and, if necessary,the dosing system) need to be connected and the device is ready for use
immediately. No complex installationand no conversion, but a high degreeof flexibility on site.

Accessories

TR 3001 TRISON RackBasket with connectors for up to8 rinsable MIS instruments. Rightand left-sided versions available.

TR 4000 TRISON Rack
For Hugo™ RAS System or Versius®
Surgical Robotic System robotic
instruments.

K 29 EM Insert basket
For holding the objects to be cleaned(e.g., standard instruments).

KT 3000 Z basket holder
For holding the insert basket.

TX 4000 Xi spacer
Spacers of the TRISON Twist Xi
for stapler instruments.

The TRISON integrated into a stainless steel row of units.

10The SONOBOARD variant of theTRISON.

D 4000 A lid
Can be used universally for all TRISON
variants. Right- and left-sided versions
available.

FT 42 foil test frame
The foil test is a simple method fordisplaying the intensity and distribution ofultrasonic cavitation in an ultrasonic bath.

11

Made in Germany
BANDELIN electronic
GmbH & Co. KG
Heinrichstraße 3 – 412207 Berlin
GERMANY
+49 30 76880-0
+49 30 7734699
info@bandelin.com
Certified toISO 9001 and ISO 13485

We will be happy to advise you personally!
Ask our experts.

Florian Knuth

Sales Manager, Medical Ultrasound +49 30 76880-212
florian.knuth@bandelin.com

The product names da Vinci, da Vinci Xi and da Vinci Si,
as well as da Vinci EndoWrist, are trademarks or registeredtrademarks of Intuitive Surgical, or its prevailing owners.
The product name Hugo™ RAS is a trademark or registeredtrademark of Covidien LP, or its prevailing owners. The
product name Versius is a trademark or registered trademark of CMR Surgical Limited, or its prevailing owners.

bandelin.com

6932-002 en/2023-10
Subject to technical changes. Illustrations are exemplaryand not to scale. Subject to change and errors excepted.
Our general terms and conditions apply.
Photos in part from: www.der-gottwald.de.