HR-AZ / HR-A
HR-250AZ
HR-251AZ
HR-150AZ
HR-100AZ

HR-250A
HR-251A
HR-150A
HR-100A

Quick Start Guide
Analytical Balance

English (EN)Guide de démarrage rapide
Balance analytique

Français (FR)

Schnellstartanleitung
Analytische Ausgeglichenheit

Deutsch (DE)

Guía de inicio rápido
Balance analítico

Español (ES)Guida introduttiva
Bilancio analitico

Italiano(IT)

快速入門指南
分析天平

中文

(CT)

1WMPD4004338

This quick start guide describes the basic functions and operations of the balance( HR-AZ / HR-A series). The functions and operating procedures of the balance are
explained in details in the "HR-AZ / HR-A series Analytical Balance INSTRUCTION
MANUAL". This manual can be downloaded from our website https://www.aandd.jp.
2021 A&D Company, Limited. All rights reserved.
No part of this publication may be reproduced, transmitted, transcribed, or translated
into any language in any form by any means without the written permission of A&D
Company, Limited.
The contents of this manual and the specifications of the instrument covered by thismanual are subject to change for improvement without notice.
Other trademarks and trade names are those of their respective owners.

The balance is a precision instrument. Unpack it carefully. It
is recommended to keep the packing materials to be used fortransporting the balance in the future.
The packing contents differ depending on the balance model.
See the following illustrations to confirm that everything is
included.

Cautions:

Verify that the AC adapter type is correct for your local voltage and receptacle type.
Use the dedicated AC adapter specified for the balance.
Do not use the AC adapter provided with the balance for other instruments.
If you use a wrong AC adapter, the balance and other equipment may not operateproperly.

2

English (EN)Caution

Do not install the balance where flammable or corrosive gas is present.

To obtain the optimum performance from the balance and acquire accurate weighing data,
note the following:
Install the balance in an environment where the temperature and humidity are notexcessive. The best operating temperature is about 20 °C ±2 °C at about 45% to 60%
RH relative humidity.
Install the balance in a location that it is free of dust.
The weighing table should be solid. (An anti-vibration table or stone table is ideal)
Place the balance on a horizontal table, and make sure that it is not tilted.
Install the balance in a stable place avoiding vibration and shock. Corners of rooms on
the first floor are best, as they are less prone to vibration.
Install the balance where it is not affected by heaters or air conditioners.
Install the balance where it is not exposed to direct sunlight.
Install the balance away from equipment that produces magnetic fields.
Be sure to connect the AC adapter to the balance and the power supply before use forat least an hour, warm up the balance.
Perform the sensitivity adjustment before using it for the first time or after having
moved it to another location so that accurate weighing can be performed.
Level the balance by adjusting the leveling
The balance is level.
feet so that the bubble of the bubble spirit
Correct weighing.
level is centered in the red circle.
The balance is tilted.
Inaccurate weighing.

Up
Down

Leveling footBubble spirit level
When the bubble is off center to the left
Turn the leveling foot on the front right inthe clockwise direction.

Bubble spirit level
Down
Up
Leveling footWhen the bubble is off center to the right
Turn the leveling foot on the front left in theclockwise direction.

When the bubble is off center to the rear When the bubble is off center to the front
Turn both leveling feet on the front in theTurn both leveling feet on the front in theclockwise direction at the same time.
counter clockwise direction at the same time.

English (EN)3

Number of statistical data (Statistical calculation mode)
The ratio between the weighing value and the weighing capacity, in a percentage, inthe weighing mode. (Capacity indicator)
Parameters of function.
Processing indicator

Response indicators

Stabilization indicator

Comparator indicators
Hold mode indicator
Units

Standby indicator
Weighing value, stored data, and items

Interval output mode.
Standby indicator.

Processing indicator
Blinking

Internal output mode.
Active indicator.

The behavior of the balance differs depending on whether you "Press and release thekey immediately" or "Press and hold the key (for 2 seconds)".
The normal weighing operation is to ”Press and release the key immediately”. Do nothold down the key unless you need to.
Press the key (Press and release the key immediately.) ..................................
Press and hold the key (for approx. 2 seconds) ..............................................
Key

When pressed
When pressed and held (for 2 seconds)
The balance is able to be turned on or off.
The standby indicator is displayed when the display is turned off.
The weighing mode is enabled when the display is turned on.
A password is required when the balance is turned on using the password
function.
This key is always available. If this key is pressed during operation, the display
will be turned off.
In the weighing mode, readabilitychanges.
The balance enters the function table
In the counting or percent mode, the mode.
balance enters the storing mode.
The weighing units selected on the
The response adjustment mode isfunction table are switched in order. performed.

4

English (EN)Key

When pressed
When pressed and held (for 2 seconds)
The HR-AZ series can performsensitivity adjustment using the builtThe sensitivity adjustment mode isin weight.
performed.
In the function table mode, it works
as cancel key.
The stable weighing data is outputted
No function at factory settings.
according to the function table.
"Title block" and "End block" is
In each operation procedure, it works
outputted according to GLP/GMP
as determination key.
settings on the function table.
The current measurement value is regarded as zero and zero is displayed.

This function performs the sensitivity adjustment of the HR-AZ series using the built-in
weight (the internal mass).

Procedure
Step 1.

Connect the AC adapter to the balance and warm up the balance for at least one
hour with nothing on the weighing pan.

Step 2.

Place the accessory breeze break on the balance.
Press the CAL key to display.

Step 3.

The sensitivity adjustment is performed using the internal mass.
Do not allow vibration or drafts to affect the balance.

Step 4.

If you choose the parameter to use the GLP output of the function table, asensitivity adjustment report is outputted after the sensitivity adjustment.

Step 5.

The balance will automatically return to weighing mode after sensitivity
adjustment.

This function performs the sensitivity adjustment using an external weight.

Procedure
Step 1.

Connect the AC adapter to the balance and warm up the it for
at least one hour with nothing on the weighing pan.

English (EN)5

Step 2.

Press and hold the CAL key until
isdisplayed for approximately two seconds. Thenrelease the key.

Step 3.

The balance displays

Press and hold forapprox. 2 sec.

.

ReleaseIf you want to change the sensitivity adjustment
weight, press the SAMPLE key and proceed to step
4. Refer to "8. CALIBRATION" in the instruction
manual for sensitivity adjustment weights that canbe selected.
If you use the sensitivity adjustment weight value
stored in the balance, proceed to step 5.
Step 4.

New value example

Specify the weight value as follows :Zero point

SAMPLE key ...........All segments blink : Weight selection mode.
The last three digits blink : Weight value adjustment.
RE-ZERO key (+) .All segments blink : Select weight.
MODE key (-)
The last three digits blink : Weight value adjustment.
PRINT key ................Weight value is stored in memory. Data is maintainedeven if removing power supply.

CAL key .....................Operation is canceled and display returns to.

Step 5.

Confirm that there is nothing on the pan and press the PRINT.
key. The balance measures the zero point. Do not allow vibration
or drafts to affect the balance. The balance displays the sensitivity
adjustment weight value.

Step 6.

Place the displayed sensitivity adjustment weight on the pan
and press the PRINT key. The balance measures the sensitivity
adjustment weight. Do not allow vibration or drafts to affect thebalance.

Step 7.

Remove the weight from the pan after displaying

Step 8.

If the GLP output is set, "Calibration Report" is outputted.

Step 9.

The balance will automatically return to the weighing mode.

Sensitivity
adjustment weight

GLP output.

Step 10. Place the sensitivity adjustment weight on the pan and confirm that themeasurement value is within ±2 digits. If it is out of that range, check the ambientconditions and repeat from steps 2.

6

English (EN)Display

Error code

Check and solutions

Stability error"Zero display" or "sensitivity adjustment" cannot be
performed due to unstable weighing value. Check the area
EC,E11 around the weighing pan. Refer to "2.1. Precautions for
installing and weighing". Prevent vibration, drafts, static
electricity and magnetic fields from influencing the balance.
Press the CAL key to return to weighing mode.

Out of range error

The input value is out of range. Input a value within the range.

Internal mass error (Only for the HR-AZ series)

EC,E17 The mechanism of the internal mass did not work properly.
Perform the operation again.

Sensitivity adjustment weight error (Positive value)

The sensitivity adjustment weight is too heavy.
Confirmthat the weighing pan is properly installed. Confirm theEC,E20sensitivity adjustment weight value. Press the CAL key toreturn to the weighing mode.

Sensitivity adjustment weight error (Negativevalue)The sensitivity adjustment weight is too light.
EC,E21 Confirm that the weighing pan is properly installed. Confirm thesensitivity adjustment weight value. Press the CAL key toreturn to the weighing mode.

Overload error

The weighing value is beyond the weighing capacity. Remove
the sample from the pan.

Weighing pan error

The weight value is too light. Confirm that the weighing pan andpan support are properly installed. Press the ON:OFF key twotimes to return to the weighing mode. If the error still persists,
perform the sensitivity adjustment.

Sample mass error

The sample unit mass value for the counting mode or thepercent mode cannot be stored because of too light. Use asample that is heavier.

Unit mass error

The sample unit mass value for the counting mode is too light.
If unit mass value is saved as is, it may cause a counting error.
Add samples of the specified number and press the PRINT key
to acquire accurate weighing.

Balance internal error

If this error appears persistently, contact the local A&D dealer.
English (EN)7

Display

Error code

Check and solutions

Battery error of clock

The back-up battery power for the clock has been completely
depleted. Adjust the date and time after pressing one of thekeys. If connecting the balance to power supply, even if the
back-up battery power for the clock has been depleted thebalance performance will be maintained. When this error
appears frequently, repair is required.

Low battery error

Voltage of the power supply has dropped.
Confirm whether the correct AC adapter is used.
If a rechargeable battery is used, replace it.

Communications error

EC,E00 A protocol error occurred in communications.
Confirm the format, baud rate and parity.
EC,E01

Undefined command error

An undefined command was received. Confirm the command.

Not readyA received command cannot be processed.
Check the timing of the transmitted command.
EC,E02 e.g. The balance received a Q command in other than theweighing mode.
e.g. The balance received a Q command while processing a
RE-ZERO command.

Timeout error

EC,E03

Whenis set, the balance did not receive the next
character of a command within the time limit of one second.
Confirm the communication.

Excess characters error

EC,E04 The balance received excessive characters in a command.
Confirm the command.

Format error

A command includes incorrect data.
EC,E06 Confirm the command.
e.g. The numeric data is incorrect.
e.g. The alphanumeric data is incorrect.

Parameter setting error

Other errors

EC,E07 The received data exceeds the acceptable range.
Confirm the parameter range of the command.
If the errors described above cannot be resolved or other errors
are displayed, contact your local A&D dealer.

8

English (EN)Weighing capacity

HR-100AZ HR-150AZ
HR-100A
HR-150A
102 g
152 g

Maximum display

102.0084 g 152.0084 g

0.1 mgLinearity±0.2 mg

HR-251AZ
HR-251A

252.008 g
62.0009 g 1
1g
0.1 mg0.1 g
0.5 mg
0.2 mg (200 to 250 g)
0.1 mg (0 to 200 g)
0.1 mg±1 mg±0.3 mg±0.3 mg

Readability
Repeatability(Standard deviation)

HR-250AZ
HR-250A
252 g
252.0084 g

1 The balance allows weighing using a precision range, even with a heavy tare placedon the pan. (Smart range function)

English (EN)9

Ce guide de démarrage rapide décrit les fonctions et les opérations de base de labalance (séries HR-AZ / HR-A). Les fonctions et procédures de fonctionnement de labalance sont expliquées en détail dans le « HR-AZ / HR-A series Analytical Balance
INSTRUCTION MANUAL (MANUEL D'INSTRUCTIONS de la balance analytiqueséries HR-AZ / HR-A) ». Ce manuel peut être téléchargé depuis notre site web
https://www.aandd.jp.
2021 A&D Company, Limited. Tous droits réservés.
La reproduction, la transmission, la transcription ou la traduction de la présente
publication, en totalité ou en partie, dans toute langue, sous quelque forme ou par
quelque procédé que ce soit est interdite sans l'autorisation écrite d'A&D Company,
Limited.
Le contenu du présent manuel et les spécifications relatives à l'instrument dont ce
manuel fait l'objet sont sujets à une modification sans préavis à des finsd'amélioration.
L'intégralité des autres marques de commerce et noms commerciaux sont la propriétéde leur détenteur respectif.

La balance est un instrument de précision. La déballer avec
soin. Il est recommandé de conserver les matériaux
d'emballage en cas de transport ultérieur de la balance.
Le contenu de l'emballage varie selon le modèle de labalance. Se référer aux illustrations suivantes pour confirmer
que tous les éléments sont inclus.

Précautions :S'assurer que le type d'adaptateur CA convient à la tension locale et au type de prise.
Utiliser l'adaptateur CA dédié spécifié pour la balance.
Ne pas utiliser l'adaptateur CA fourni avec la balance pour les autres instruments.
Si un adaptateur CA incorrect est utilisé, la balance et les autres équipements peuvent
ne pas fonctionner correctement.
10

Français (FR)Attention Ne pas installer la balance dans un environnement où des gaz inflammables
ou corrosifs sont présents.
Pour obtenir des performances optimales de la balance et des données de pesage
précises, noter ce qui suit :Installer la balance dans un environnement où la température et l'humidité ne sont pas
excessives. La meilleure température de fonctionnement est d'environ 20 °C ±2 °C
avec environ 45 à 60 % d'humidité relative.
Installer la balance dans un endroit exempt de poussière.
La table de pesage doit être robuste. (Une table anti-vibrations ou une table en pierre
est idéale)Placer la balance sur une table horizontale et s'assurer qu'elle n'est pas inclinée.
Installer la balance dans un endroit stable, exempt de vibrations et de chocs. Les coins
des pièces au rez-de-chaussée sont idéaux car ils sont moins exposés aux vibrations.
Installer la balance dans un endroit non affecté par des systèmes de chauffage ou declimatisation.
Installer la balance dans un endroit non exposé à la lumière directe du soleil.
Installer la balance à l'écart d'équipements produisant des champs magnétiques.
Veiller à brancher l'adaptateur CA à la balance et l'alimentation électrique avant
utilisation pendant au moins une heure afin de réchauffer la balance.
Effectuer le réglage de la sensibilité avant utilisation de la balance pour la première fois ou
après l'avoir déplacée vers un autre endroit afin de pouvoir effectuer un pesage précis.
Mettre la balance à niveau en réglant les
La balance est de niveau.
pieds réglables de manière à ce que laPesage correct.
bulle du niveau à bulle soit centrée dans le
La balance est inclinée.
cercle rouge.
Pesage inexact.
Haut
Bas

Pied réglable

Niveau à bulle

Niveau à bulle

Bas
Haut
Pied réglable

Quand la bulle dévie vers la gauche
Tourner le pied réglable avant droit dansle sens des aiguilles d'une montre.

Quand la bulle dévie vers la droite
Tourner le pied réglable avant gauche dansle sens des aiguilles d'une montre.

Quand la bulle dévie vers l'arrière
Tourner simultanément les deux pieds
réglables avant dans le sens des aiguilles
d'une montre.

Quand la bulle dévie vers l'avant
Tourner simultanément les deux pieds
réglables avant dans le sens contraire desaiguilles d'une montre.

Français (FR)11

Nombre de données statistiques (Mode de calcul statistique)
Le rapport entre la valeur de pesage et la capacité de pesage, en pourcentage, dansle mode de pesage. (Indicateur de capacité)
Paramètres de la fonction.
Indicateur de traitement

Indicateurs de réponse

Indicateur de stabilisation

Indicateurs de comparateur
Indicateur du mode de maintien
UnitésIndicateur de veille
Valeur de pesage, données enregistrées et éléments

Indicateur d'exportation de l'intervalle.
Indicateur de veille.

Indicateur de traitement
Clignotant

Indicateur de sortie interne.
Indicateur d'activité.

Le comportement de la balance diffère selon s'il faut « Appuyer et relâcher la touche
immédiatement » ou « Appuyer sur la touche et la maintenir enfoncée (pendant 2
secondes) ».
L'opération de pesage normale consiste à « Appuyer et relâcher la touche
immédiatement ». Ne pas maintenir enfoncée la touche sauf si cela est nécessaire.
Appuyer sur la touche (Appuyer et relâcher la touche immédiatement). ........
Appuyer sur la touche et la maintenir enfoncée (pendant environ 2 secondes) .........
ToucheEnfoncée et maintenue enfoncée(pendant 2 secondes)

EnfoncéeLa balance peut être allumée ou éteinte.
L'indicateur de veille s'affiche lorsque l'écran est mis hors tension.
Le mode de pesage est activé lorsque l'écran est mis sous tension.
Un mot de passe est nécessaire lorsque la balance est allumée en utilisant lafonction de mot de passe.
Cette touche est toujours disponible. Si cette touche est enfoncée pendantl'opération, l'écran s'éteint.
En mode de pesage, la lisibilité
change.
La balance entre en mode table deEn mode de comptage ou defonctions.
pourcentage, la balance entre enmode d'enregistrement.

12

Français (FR)

ToucheEnfoncée et maintenue enfoncée(pendant 2 secondes)

EnfoncéeLes unités de pesage sélectionnées
Le mode de réglage de la réponse est
dans la table de fonctions sontexécuté.
basculées dans l'ordre.
La série HR-AZ peut effectuer unréglage de la sensibilité en utilisant lepoids intégré.
Le mode de réglage de la sensibilité est
En mode table de fonctions, cela
exécuté.
fonctionne comme une touched'annulation.
Aucune fonction attribuée comme
Les données de pesage stables sontparamètre d'usine.
émises selon la table de fonctions.
Le « Title block (Bloc de titre) » et leDans chaque procédure d'opération,« End block (Bloc de fin) » sont émiscela fonctionne comme une touche
selon les paramètres GLP/GMP dede détermination.
la table de fonctions.
La valeur de mesure actuelle est considérée comme nulle et le zéro est affiché.

Cette fonction permet de régler la sensibilité de la série HR-AZ en utilisant le poids
intégré (la masse interne).

Procédure
Étape 1.

Raccorder l'adaptateur CA à la balance et laisser la balance chauffer pendant
au moins une heure, plateau de pesage vide.

Étape 2.

Placer l'accessoire coupe-brise sur la balance.
Appuyer sur la touche CAL pour afficher

Étape 3.

Le réglage de la sensibilité est effectué à l'aide de la masse interne.
Éviter d'exposer la balance à des vibrations ou des courants d'air.

Étape 4.

En choisissant le paramètre qui permet d'utiliser la sortie GLP de la table defonctions, un rapport du réglage de la sensibilité est émis après le réglage de lasensibilité.

Étape 5.

La balance retourne automatiquement en mode de pesage après le réglage dela sensibilité.

Français (FR)13.

La fonction effectue le réglage de la sensibilité à l'aide d'un poids externe.

Procédure
Étape 1.

Raccorder l'adaptateur CA à la balance et la laisser chauffer
pendant au moins une heure, plateau de pesage vide.

Étape 2.

Appuyer sur la touche CAL et la maintenir en
foncée jusqu'à ce que
s'affiche
pendant environ deux secondes. Ensuite,
relâcher la touche.

Étape 3.

La balance afficheAppuyer sur la touche et lamaintenir enfoncée pendantenviron 2 sec.
Relâcher.

Si le poids de réglage de la sensibilité doit êtrechangé, appuyer sur la touche SAMPLE etpasser à l'étape 4. Se référer à « 8.
CALIBRATION (8. ÉTALONNAGE) » dans lemanuel d'instructions pour les poids de réglage
de la sensibilité qui peuvent être sélectionnés.
Si la valeur du poids de réglage de la sensibilité
enregistré dans la balance est utilisé, passer àl'étape 5.

Exemple de nouvelle valeur

Étape 4. Spécifier la valeur du poids comme suit :Touche SAMPLE ... Tous les segments clignotent : Mode de sélection dupoids.
Les trois derniers chiffres clignotent : Réglage de lavaleur du poids.
Touche RE-ZERO (+) .. Tous les segments clignotent : Sélectionner un poids.
Touche MODE (-)
Les trois derniers chiffres clignotent : Réglage de lavaleur du poids.
Touche PRINT ...... La valeur du poids est enregistrée dans la mémoire.
Les données sont conservées même si l'alimentationélectrique est retirée.

Point zéro

Poids du réglage
de la sensibilité

Touche CAL .......... L'opération est annulée et l'affichage revient à.
Étape 5. Confirmer qu'il n'y a rien sur le plateau et appuyer sur la touche
PRINT . La balance mesure le point zéro. Éviter d'exposer labalance à des vibrations ou des courants d'air. La balance
affiche la valeur du poids de réglage de la sensibilité.
Étape 6. Placer le poids de réglage de la sensibilité sur le plateau etappuyer sur la touche PRINT . La balance mesure le poids deréglage de la sensibilité. Éviter d'exposer la balance à desvibrations ou des courants d'air.
Étape 7. Retirer le poids du plateau après que

Exportation GLP

est affiché.

Étape 8. Si la sortie GLP est définie, le "rapport d'étalonnage" est émis.
14

Français (FR)

Étape 9. La balance retourne automatiquement en mode de pesage.
Étape 10. Placer le poids de réglage de la sensibilité sur le plateau et confirmer que lavaleur de mesure se situe dans une limite de ±2 chiffres. Si elle se situe endehors de cette plage, vérifier les conditions ambiantes et recommencer à partir
de l'étape 2.

Affichage

Code d'erreur

Vérification et solutions

Erreur de stabilité

L'« affichage du zéro » ou le « réglage de la sensibilité » nepeut être effectué en raison d'une valeur de pesage instable.
Vérifier la zone autour du plateau de pesage. Se référer à « 2.1.
EC,E11 Précautions à prendre pour l'installation et le pesage ». Éviter
les vibrations, les courants d'air, l'électricité statique et leschamps magnétiques qui peuvent nuire au fonctionnement de labalance.
Appuyer sur la touche CAL pour revenir en mode de pesage.

Erreur de dépassement de plage

La valeur d'entrée se situe hors de la plage. Entrer une valeurdans la plage.

Erreur de masse interne (uniquement pour la série
HR-AZ)

EC,E17 Le mécanisme de la masse interne n'a pas fonctionné
correctement. Effectuer à nouveau l'opération.

Erreur de poids du réglage de la sensibilité (valeur
positive)

Le poids du réglage de la sensibilité est trop lourd.
EC,E20 Confirmer que le plateau de pesage est correctement installé.
Vérifier la valeur du poids de réglage de la sensibilité. Appuyer
sur la touche CAL pour revenir en mode de pesage.

Erreur de poids du réglage de la sensibilité (valeurnégative)

Le poids du réglage de la sensibilité est trop léger.
EC,E21 Confirmer que le plateau de pesage est correctement installé.
Vérifier la valeur du poids de réglage de la sensibilité. Appuyer
sur la touche CAL pour revenir en mode de pesage.

Erreur de surcharge

La valeur de pesage se situe au-delà de la capacité de labalance. Retirer l'échantillon du plateau.

Français (FR)15

Affichage

Code d'erreur

Vérification et solutions

Erreur du plateau de pesage

La valeur du poids est trop légère. Confirmer que le plateau depesage et le support de plateau sont correctement installés.
Appuyer sur la touche ON:OFF deux fois pour revenir en modede pesage. Si l'erreur persiste, effectuer le réglage de lasensibilité.

Erreur de masse de l'échantillon

La valeur de la masse de l'unité d'échantillonnage pour le modede comptage ou le mode de pourcentage ne peut pas êtreenregistrée car elle est trop légère. Utiliser un échantillon qui
est plus lourd.

Erreur de la masse unitaire

La valeur de la masse de l'unité d'échantillonnage en mode decomptage est trop légère.
Si la valeur de la masse unitaire est enregistrée telle quelle,cela peut provoquer une erreur de comptage.
Ajouter des échantillons pour le nombre spécifié et appuyer sur
la touche PRINT pour obtenir un pesage précis.

Erreur interne de la balance

Si cette erreur s'affiche en permanence, contacter le distributeur
A&D local.

Erreur de batterie de l'horloge

L'alimentation de la batterie de secours de l'horloge est
complètement épuisée. Régler la date et l'heure après avoir
appuyé sur l'une des touches. En raccordant la balance àl'alimentation électrique, même si la batterie de secours del'horloge est épuisée, les performances de la balance seront
maintenues. Lorsque cette erreur apparaît fréquemment, uneréparation est nécessaire.

Erreur de batterie faible

La tension de l'alimentation électrique a baissé.
Confirmer que l'adaptateur CA utilisé est correct.
Si une batterie rechargeable est utilisée, la remplacer.

Erreur de communications

EC,E00 Une erreur de protocole est survenue dans les communications.
Confirmer le format, la vitesse de transmission et la parité.
EC,E01

Erreur de commande non définie

Une commande indéfinie a été reçue. Confirmer la commande.

Non prête

Une commande reçue ne peut pas être traitée.
Vérifier le timing de la commande transmise.
EC,E02 Par ex. La balance a reçu une commande Q dans un autre
mode que celui de pesage.
Par ex. La balance a reçu une commande Q pendant letraitement d'une commande RE-ZERO.
16

Français (FR)Affichage

Code d'erreur
EC,E03

Vérification et solutions

Erreur de délai d'expirationSi
est réglé, la balance n'a pas reçu le caractère
suivant d'une commande dans la limite d'une seconde.
Confirmer la communication.

Erreur de caractères excessifs
EC,E04La balance a reçu des caractères excessifs dans une
commande.
Confirmer la commande.

Erreur de format

Une commande inclut des données incorrectes.
EC,E06 Confirmer la commande.
Par ex. Les données numériques sont incorrectes.
Par ex. Les données alphanumériques sont incorrectes.

Erreur de réglage des paramètres

EC,E07 Les données reçues dépassent la plage acceptable.
Confirmer la plage des paramètres de la commande.
Si les erreurs décrites ci-dessus ne peuvent être résolues ou si
Autres erreursd'autres erreurs sont affichées, contacter votre revendeur A&D
local.

Capacité de pesage

HR-100AZ
HR-100A
102 g

HR-150AZ
HR-150A
152 g

Affichage maximum

102,0084 g 152,0084 g

0,1 mg

Linéarité±0,2 mg

HR-251AZ
HR-251A

252,008 g
62,0009 g 1
1g
0,1 mg0,1 g
0,5 mg0,2 mg (200 à 250 g)
0,1 mg (0 à 200 g)
0,1 mg±1 mg±0,3 mg±0,3 mg

Lisibilité
Répétabilité(Écart type)

HR-250AZ
HR-250A
252 g
252,0084 g

1 La balance permet de peser à l'aide d'une plage de précision, même avec une tarelourde placée sur le plateau. (Fonction de plage intelligente)

Français (FR)17

In dieser Schnellstartanleitung werden die Grundfunktionen und die Bedienung derWaage beschrieben (Serie HR-AZ / HR-A). Die Funktionen und die
Bedienungsverfahren der Waage werden im „HR-AZ / HR-A series Analytical
Balance INSTRUCTION MANUAL (Analysenwaagen-BEDIENUNGSHANDBUCH
der Serie HR-AZ / HR-A)“ näher erklärt. Dieses Handbuch kann von unserer
Webseite https://www.aandd.jp heruntergeladen werden.
2021 A&D Company, Limited. Alle Rechte vorbehalten.
Kein Teil dieser Publikation darf ohne schriftliche Genehmigung der A&D Company,
Limited in irgendeiner Form vervielfältigt, übertragen, umgeschrieben oder in eine
andere Sprache übersetzt werden.
Der Inhalt dieses Handbuchs und die Spezifikationen des in diesem Handbuch
behandelten Geräts können zum Zwecke der Verbesserung ohne vorherige
Ankündigung geändert werden.
Andere Warenzeichen und Handelsnamen sind Eigentum der jeweiligen Inhaber.

Diese Waage ist ein Präzisionsinstrument. Gehen Sie beim
Auspacken vorsichtig vor. Es wird empfohlen, die
Verpackungsmaterialien für eventuelle zukünftige Transporte
aufzubewahren.
Der Inhalt der Verpackung ist je nach Modell der Waage
unterschiedlich. Überprüfen Sie anhand der folgenden
Abbildungen, ob alle Teile enthalten sind.

Vorsichtshinweise:Überprüfen Sie, dass der Wechselstromadapter der Spannung und dem
Steckdosentyp vor Ort entspricht.
Verwenden Sie den für die Waage angegebenen Wechselstromadapter.
Verwenden Sie den mit der Waage mitgelieferten Wechselstromadapter nicht für
andere Geräte.
Wenn Sie einen falschen Wechselstromadapter verwenden, funktionieren die Waage
und andere Geräte eventuell nicht richtig.

18

Deutsch (DE)Vorsicht

Stellen Sie die Waage nicht in einer Umgebung auf, in der entzündliche oderätzende Gase auftreten.

Um die Leistungsfähigkeit der Waage zu optimieren und exakte Wägedaten zu erhalten,
beachten Sie bitte folgende Hinweise:
Stellen Sie die Waage in einer Umgebung ohne extreme Temperaturen und
Feuchtigkeit auf. Die optimale Betriebstemperatur beträgt etwa 20 °C ±2 °C bei einer
relativen Luftfeuchtigkeit von etwa 45 % bis 60 % RH.
Stellen Sie die Waage an einem staubfreien Ort auf.
Der Wägetisch sollte stabil sein. (Ideal ist ein Anti-Vibrationstisch oder ein Steintisch.)
Platzieren Sie die Waage auf einem horizontalen Tisch und stellen Sie sicher, dass ernicht geneigt ist.
Stellen Sie die Waage auf eine stabile Fläche, die weder Vibrationen noch Stößen
ausgesetzt ist. Am besten geeignet sind die Eckräume im Erdgeschoss, weil sie
weniger Vibrationen ausgesetzt sind.
Installieren Sie die Waage an einem Ort, wo sie nicht durch Heizkörper oder
Klimaanlagen beeinträchtigt wird.
Installieren Sie die Waage an einem Ort, wo sie keiner direkten Sonneneinstrahlung
ausgesetzt ist.
Stellen Sie die Waage nicht in der Nähe von Geräten auf, die magnetische Felder
erzeugen.
Achten Sie darauf, den Wechselstromadapter vor der Verwendung mindestens eine
Stunde lang mit der Waage und der Stromversorgung zu verbinden und die Waage
aufzuwärmen.
Führen Sie vor dem ersten Gebrauch der Waage oder nachdem sie an einen anderen
Ort gestellt wurde die Empfindlichkeitsanpassung aus, damit ein genaues Wiegenerzielt werden kann.
Nivellieren Sie die Waage aus, indem
Die Waage ist optimal ausnivelliert.
Sie die Nivellierungsfüße anpassen,
Korrektes Wiegen.
sodass sich die Blase der Libelle in der
Die Waage ist geneigt.
Wasserwaage im roten Kreis befindet.
Inkorrektes Wiegen.
Nach oben
Nach unten

Nivellierungsfuß

Wasserwaage

Wasserwaage
Nach unten
Nach oben
Nivellierungsfuß

Wenn sich die Libelle links von der Mitte
befindet
Drehen Sie den Nivellierungsfuß vorn
rechts im Uhrzeigersinn.

Wenn sich die Libelle rechts von der Mitte
befindet
Drehen Sie den Nivellierungsfuß vorn links
im Uhrzeigersinn.

Deutsch (DE)19

Wenn sich die Libelle hinter der Mitte
befindet
Drehen Sie beide Nivellierungsfüße an der
Vorderseite gleichzeitig im Uhrzeigersinn.

Wenn sich die Libelle vor der Mitte
befindet
Drehen Sie beide Nivellierungsfüße an der
Vorderseite gleichzeitig gegen den
Uhrzeigersinn.

Anzahl der statistischen Daten (Statistik-Berechnungsmodus)
Das Verhältnis zwischen dem Wägewert und der Wägekapazität, in Prozent, im
Wägemodus. (Kapazitätsanzeige)
Parameter der Funktion.
Verarbeitungsanzeige

Reaktionsanzeigen

Stabilisierungsanzeige

Vergleichsanzeigen
Hold-Modus-Anzeige
Einheiten

Bereitschaftsanzeige
Wägewert, gespeicherte Daten und Elemente
Verarbeitungsanzeige
BlinktIntervall-Ausgabemodus.
Bereitschaftsanzeige.

Interner Ausgabemodus.
Aktiv-Anzeige.

Das Verhalten der Waage unterscheidet sich je nachdem, ob Sie die „Taste drücken undsofort wieder loslassen“ oder die „Taste drücken und 2 Sekunden lang gedrückt halten“.
Der normale Wägevorgang ist „Taste drücken und sofort wieder loslassen“. Halten Sie
die Taste nur gedrückt, wenn es notwendig ist.
Taste drücken (Taste drücken und sofort wieder loslassen) ............................
Taste drücken und gedrückt halten (ca. 2 Sekunden lang) ...........................
Taste

Gedrückt und gedrückt gehalten(für 2 Sekunden)
Die Waage kann ein- und ausgeschaltet werden.
Wenn die Anzeige ausgeschaltet ist, erscheint die Bereitschaftsanzeige.
Der Wägemodus wird mit dem Einschalten der Anzeige aktiviert.
Ein Passwort wird benötigt, wenn die Waage unter Verwendung derPasswortfunktion eingeschaltet wird.
Diese Taste ist immer verfügbar. Wenn diese Taste während des Betriebs
gedrückt wird, wird die Anzeige ausgeschaltet.
20
Deutsch (DE)
GedrücktTasteGedrückt und gedrückt gehalten(für 2 Sekunden)

Gedrückt

Im Wägemodus ändert sich die
Lesbarkeit.
Die Waage wechselt in den
Im Zähl- und Prozentmodus wechselt
Funktionstabellen-Modus.
die Waage in den Speichermodus.
Die in der Funktionstabelle
Der Reaktionsanpassungsmodus
ausgewählten Wägeeinheiten werden
wird ausgeführt.
der Reihe nach umgeschaltet.
Die Serie HR-AZ kann die
Empfindlichkeitsanpassung mithilfe des Der
eingebauten Gewichtes ausführen.
Empfindlichkeitsanpassungsmodus
Im Funktionstabellen-Modus funktioniert wird ausgeführt.
sie als Taste zum Abbrechen.
Bei Werkseinstellung keine
Die stabilen Wägedaten werden gemäß
Funktion.
der Funktionstabelle ausgegeben.
„Title block (Titelblock)“ und „End
Sie funktioniert in allen
block (Endblock)“ werden gemäß
Bedienungsverfahren als Taste zum
den GLP/GMP-Einstellungen in
Festlegen.
der Funktionstabelle ausgegeben.
Der aktuelle Messwert wird als Null angesehen und Null wird angezeigt.

Diese Funktion führt die Empfindlichkeitseinstellung bei der Serie HR-AZ mithilfe deseingebauten Gewichtes (der internen Masse) aus.

Vorgang

Schritt 1. Schließen Sie den Wechselstromadapter an die Waage an und lassen Sie die
Waage mindestens 1 Stunde lang aufwärmen, ohne etwas in die Waagschalezu legen.
Schritt 2. Stellen Sie den Windschutz (Zubehör) auf die Waage.
Drücken Sie die CAL -Taste, um
anzuzeigen.
Schritt 3. Die Empfindlichkeitsanpassung wird mithilfe der internen Masse durchgeführt.
Sorgen Sie dafür, dass die Waage weder Vibrationen noch Zugluft ausgesetzt ist.
Schritt 4. Wenn Sie den Parameter zur Verwendung der GLP-Ausgabe der
Funktionstabelle wählen, wird nach der Empfindlichkeitsanpassung ein Berichtüber die Empfindlichkeitsanpassung ausgegeben.
Schritt 5. Die Waage kehrt nach der Empfindlichkeitsanpassung automatisch wieder in
den Wägemodus zurück.
Deutsch (DE)21

Diese Funktion führt die Empfindlichkeitsanpassung mithilfe eines externen Gewichts aus.

Vorgang

Schritt 1. Schließen Sie den Wechselstromadapter an die
Waage an und lassen Sie die Waage mindestens 1
Stunde lang aufwärmen, ohne etwas in die
Drücken und ca. 2 Sekunden
Waagschale zu legen.
lang gedrückt halten.
Schritt 2. Taste CAL drücken und gedrückt halten, bis
ca. zwei Sekunden lang angezeigt wird.
Lassen Sie dann die Taste wieder los.
Schritt 3. Die Waage zeigt

Loslassen

an.

Wenn Sie das Empfindlichkeitsanpassungsgewichtändern möchten, drücken Sie die SAMPLE -Tasteund gehen Sie zu Schritt 4. Für Details zu
auswählbaren
Empfindlichkeitsanpassungsgewichten siehe „8.
CALIBRATION (8. KALIBRIERUNG)“ im
Bedienungshandbuch.

Beispiel für neuen Wert

Falls Sie den in der Waage gespeicherten Wert desEmpfindlichkeitsanpassungsgewichts verwenden, fahren Sie
mit Schritt 5 fort.

Nullpunkt

Schritt 4. Legen Sie den Wert des Gewichts wie folgt fest:
SAMPLE -Taste ..... Alle Segmente blinken : Gewichts-Auswahlmodus.
Die letzten drei Ziffern blinken :Gewichtswerteinstellung.
RE-ZERO -Taste (+) ... Alle Segmente blinken : Gewicht auswählen.
MODE -Taste (-) Die letzten drei Ziffern blinken :Gewichtswerteinstellung.

PRINT -Taste ........ Gewichtswert ist im Speicher abgelegt. Die Daten

Empfindlichkeitsa
npassungswert

bleiben auch bei Unterbrechung der Stromversorgung
erhalten.

CAL -Taste ........... Der Betrieb wird abgebrochen und die Anzeige kehrt

zurück zu.

Schritt 5. Vergewissern Sie sich, dass sich nichts auf der Waagschale
befindet und drücken Sie die PRINT. -Taste. Die Waage misst
den Nullpunkt. Sorgen Sie dafür, dass die Waage weder
Vibrationen noch Zugluft ausgesetzt ist. Die Waage zeigt den
Wert des Empfindlichkeitsanpassungsgewichts an.

GLP-Ausgabe

Schritt 6. Legen Sie das angezeigte Empfindlichkeitsanpassungsgewicht
auf die Waagschale und drücken Sie die PRINT -Taste. Die Waage
22

Deutsch (DE)misst das Empfindlichkeitsanpassungsgewicht. Sorgen Sie dafür,dass die Waage weder Vibrationen noch Zugluft ausgesetzt ist.
Schritt 7. Nehmen Sie das Gewicht nach der Anzeige vonvon der Waagschale.

Schritt 8. Falls die GLP-Ausgabe aktiviert ist, wird der „Calibration Report(Kalibrierungsbericht)“ ausgegeben.
Schritt 9. Die Waage kehrt automatisch wieder in den Wägemodus zurück.
Schritt 10. Legen Sie das Empfindlichkeitsanpassungsgewicht auf die Waagschale undbestätigen Sie, dass der Messwert im Bereich von ±2 Ziffern liegt. Liegt er
außerhalb dieses Bereichs, prüfen Sie die Umgebungsbedingungen undwiederholen Sie ab Schritt 2.

AnzeigeFehlercodeÜberprüfung und LösungenStabilitätsfehler„Nullanzeige“ oder „Empfindlichkeitsanpassung“ können
aufgrund des instabilen Gewichtswerts nicht ausgeführt werden.
Überprüfen Sie den Bereich rund um die Waagschale. Siehe
EC,E11 „2.1. Vorsichtsmaßnahmen zum Installieren und Wiegen“.
Vermeiden Sie die Beeinflussung der Waage durch Vibration,
Zugluft, statische Elektrizität und Magnetfelder. Drücken Sie die
CAL -Taste, um zum Wägemodus zurückzukehren.

Fehler „Außerhalb des Bereichs“

Der Eingabewert liegt außerhalb des Bereichs. Geben Sie einen
Wert innerhalb des Bereichs ein.

Fehler mit der internen Masse (Nur bei der Serie HR-AZ)
EC,E17

Der Mechanismus der internen Masse funktioniert nicht korrekt.
Führen Sie den Vorgang erneut durch.

Fehler des Empfindlichkeitsanpassungsgewichts(Positiver Wert)

Das Empfindlichkeitsanpassungsgewicht ist zu schwer.
Vergewissern Sie sich, dass die Waagschale korrekt installiert
EC,E20wurde. Bestätigen Sie den Wert desEmpfindlichkeitsanpassungsgewichts. Drücken Sie die CAL Taste, um zum Wägemodus zurückzukehren.

Fehler des Empfindlichkeitsanpassungsgewichts(Negativer Wert)

Das Empfindlichkeitsanpassungsgewicht ist zu leicht.
Vergewissern Sie sich, dass die Waagschale korrekt installiert
EC,E21
wurde. Bestätigen Sie den Wert desEmpfindlichkeitsanpassungsgewichts. Drücken Sie die CAL Taste, um zum Wägemodus zurückzukehren.

Deutsch (DE)23

Anzeige

FehlercodeÜberlastfehlerÜberprüfung und Lösungen

Der Wägewert hat die Wägekapazität überschritten. Nehmen
Sie die Probe von der Waagschale.

Waagschalen-Fehler

Der Gewichtswert ist zu leicht. Vergewissern Sie sich, dass die
Waagschale und die Schalenhalterung ordnungsgemäß
installiert sind. Drücken Sie die ON:OFF -Taste zwei Mal, um
zum Wägemodus zurückzukehren. Wenn der Fehler weiterhin
besteht, führen Sie die Empfindlichkeitsanpassung durch.

Probenmassen-Fehler

Der Probeneinheitsmassenwert für den Zählmodus oder den
Prozentmodus kann nicht gespeichert werden, weil er zu gering
ist. Verwenden Sie eine schwerere Probe.

Einheitsmasse-Fehler

Der Probeneinheitsmassenwert ist für den Zählmodus zu leicht.
Wenn der Einheitsmassenwert unverändert gespeichert wird,
kann dies zu einem Zählfehler führen. Fügen Sie Proben derangegebenen Zahl hinzu und drücken Sie die PRINT -Taste, um
ein genaues Wiegen zu erzielen.

Interner Fehler der Waage

Wenn dieser Fehler immer wieder auftritt, wenden Sie sich an
den lokalen A&D-Händler.

Batteriefehler der Uhr

Die Backup-Batterie der Uhr ist vollständig entladen. Stellen Sie
das Datum und die Uhrzeit ein, nachdem Sie eine der Tasten
gedrückt haben. Wenn Sie die Waage an die Stromversorgung
anschließen, bleibt die Leistung der Waage auch dann erhalten,
wenn die Backup-Batterie für die Uhr leer ist. Wenn dieser
Fehler häufig auftritt, ist eine Reparatur erforderlich.

Fehler durch niedrigen BatteriestandDie Spannung der Stromversorgung ist gesunken.
Vergewissern Sie sich, dass der richtige Wechselstromadapter
verwendet wird. Wenn eine wiederaufladbare Batterie
verwendet wird, ersetzen Sie diese.

Kommunikationsfehler

EC,E00 Bei der Übertragung trat ein Protokollfehler auf.
Überprüfen Sie Format, Baudrate und Parität.

Fehler Undefinierter Befehl

EC,E01 Ein nicht definierter Befehl wurde empfangen. Überprüfen Sie
den Befehl.

24

Deutsch (DE)Anzeige

Fehlercode

Nicht bereitÜberprüfung und LösungenEin empfangener Befehl kann nicht verarbeitet werden.
Überprüfen Sie den Zeitpunkt des übertragenen Befehls.
EC,E02 z. B. Die Waage erhielt einen Q-Befehl in einem anderen
Modus als dem Wägemodus.
z. B. Die Waage erhielt einen Q-Befehl während derVerarbeitung eines RE-ZERO-Befehls.

Timeout-Fehler

EC,E03Wenn
eingestellt ist, hat die Waage das nächste
Zeichen eines Befehls in der vorgegebenen Frist von einer
Sekunde nicht erhalten. Überprüfen Sie die Kommunikation.

Fehler „Überschüssige Zeichen“

EC,E04 Die Waage erhielt überschüssige Zeichen in einem Befehl.
Überprüfen Sie den Befehl.

Formatfehler

EC,E06

Ein Befehl enthält falsche Daten. Überprüfen Sie den Befehl.
z. B. Die numerischen Daten sind inkorrekt.
z. B. Die alphanumerischen Daten sind inkorrekt.

Parameter-Einstellungsfehler

EC,E07 Die empfangenen Daten überschreiten den zulässigen Bereich.
Überprüfen Sie den Parameterbereich des Befehls.
Wenn die oben beschriebenen Fehler nicht behoben werden
Andere Fehler
können oder andere Fehler angezeigt werden, wenden Sie sichan Ihren A&D-Händler vor Ort.

Wägekapazität

HR-100AZ
HR-100A
102 g

HR-150AZ
HR-150A
152 g

Maximale Anzeige

102,0084 g 152,0084 g

0,1 mgLinearität±0,2 mg

HR-251AZ
HR-251A

252,008 g
62,0009 g 1
1g
0,1 mg0,1 g
0,5 mg0,2 mg (200 bis 250 g)
0,1 mg (0 bis 200 g)
0,1 mg±1 mg±0,3 mg±0,3 mg

Lesbarkeit
Wiederholbarkeit(Standardabweichung)HR-250AZ
HR-250A
252 g
252,0084 g

1 Die Waage ermöglicht das Wiegen im Präzisionsbereich, selbst bei einer schweren
auf die Waagschale gelegten Tara. (Smart-Bereichsfunktion)

Deutsch (DE)25

En esta guía de inicio rápido se describen las funciones y operaciones básicas de labáscula (series HR-AZ / HR-A). Las funciones y los procedimientos de uso de labáscula se explican detalladamente en “HR-AZ / HR-A series Analytical Balance
INSTRUCTION MANUAL” (MANUAL DE INSTRUCCIONES de las básculasanalíticas de las series HR-AZ / HR-A). Este manual se puede descargar de nuestro
sitio web https://www.aandd.jp.
2021 A&D Company, Limited. Todos los derechos reservados.
Ninguna parte de esta publicación puede ser reproducida, transmitida, transcrita o
traducida a ningún idioma ni por cualquier medio sin la autorización escrita de A&D
Company, Limited.
El contenido del presente manual y las especificaciones del instrumento cubierto por
este están sujetos a cambios para su mejora sin previo aviso.
Las demás marcas y nombres comerciales pertenecen a sus respectivos propietarios.

La báscula es un instrumento de precisión. Desembálelo con
cuidado. Se recomienda conservar los materiales deembalaje con el fin de utilizarlos para transportar la báscula
en el futuro.
El contenido del paquete varía según el modelo de labáscula. Consulte las siguientes ilustraciones para
comprobar que incluye todos los componentes.

Medidas de precaución:

Verifique que el tipo de adaptador de CA sea el correcto para la tensión que se utiliza
en su zona y para el tipo de enchufe.
Utilice el adaptador de CA específico indicado para la báscula.
No utilice el adaptador de CA suministrado con la báscula con otros instrumentos.
Si utiliza un adaptador de CA incorrecto, la báscula y el resto del equipo podrían no
funcionar bien.
26

Español (ES)Medidas de precaución No coloque la báscula en lugares donde haya gases
inflamables o corrosivos.
Para obtener un rendimiento óptimo de la báscula y adquirir datos de pesaje precisos,
tenga en cuenta lo siguiente:
Coloque la báscula en un entorno cuya temperatura y humedad no sean excesivos.
La temperatura de funcionamiento óptima es de aproximadamente 20 °C ±2 °C con
una humedad relativa de alrededor del 45-60 %.
Coloque la báscula en un lugar sin polvo.
La mesa donde se realice el pesaje debe ser firme. (Lo ideal es usar una mesa depiedra o antivibraciones)
Coloque la báscula sobre una mesa horizontal y compruebe que no esté inclinada.
Coloque la báscula en un lugar estable; evite las vibraciones y los golpes. El lugar
idóneo son las esquinas de las salas situadas en la primera planta, ya que son zonas
menos propensas a las vibraciones.
Coloque la báscula en un lugar en el que no se vea afectada por calefactores ni aires
acondicionados.
Coloque la báscula en un lugar en el que no esté expuesta a la luz solar directa.
Coloque la báscula alejada de equipos que produzcan campos magnéticos.
Asegúrese de conectar el adaptador de CA a la báscula y a la fuente de alimentación
antes de usarla durante un mínimo de una hora para calentar la báscula.
Realice el ajuste de la sensibilidad antes de utilizarla por primera vez o después dehaberla movido a otra ubicación para poder realizar un pesaje preciso.
Nivele la báscula ajustando las ruedas
La báscula está nivelada.
niveladoras para que la burbuja del nivel dePesaje correcto.
burbuja esté centrada en el círculo rojo.
La báscula está inclinada.
Pesaje inexacto.

ArribaAbajo

Rueda niveladora

Nivel de burbuja

Nivel de burbuja

Abajo
ArribaRueda niveladora

Si la burbuja está a la izquierda
Gire la rueda niveladora delantera derechaen el sentido de las agujas del reloj.

Si la burbuja está a la derecha
Gire la rueda niveladora delantera izquierda
en el sentido de las agujas del reloj.

Si la burbuja está detrás
Gire las dos ruedas niveladoras delanteras ala vez en el sentido de las agujas del reloj.

Si la burbuja está delante
Gire las dos ruedas niveladoras delanteras a lavez en el sentido contrario a las agujas del reloj.

Español (ES)27

Número de datos estadísticos (modo de cálculo estadístico)
La relación entre el valor de pesaje y la capacidad de pesaje, en porcentaje, en elmodo de pesaje. (Indicador de capacidad)
Parámetros de la función.
Indicador de procesamiento

Indicadores de respuesta

Indicador de estabilización

Indicadores del comparador
Indicador de modo de espera
UnidadesIndicador de espera
Valor de pesaje, datos almacenados y elementos

Modo de salida de intervalos.
Indicador de espera.

Indicador de procesamiento
Parpadeando

Modo de salida interna.
Indicador de activación.

El funcionamiento de la báscula varía según si se realiza “Pulsar la tecla y soltarla
inmediatamente”, o bien “Mantener pulsada la tecla (durante 2 segundos)”.
La operación de pesaje estándar se realiza al “Pulsar la tecla y soltarla
inmediatamente”. No mantenga pulsada la tecla a menos que sea necesario.
Pulsar la tecla (pulsarla y soltarla inmediatamente). ........................................
Mantener pulsada la tecla (durante aproximadamente 2 segundos) ..............
Tecla

Cuando se mantiene pulsada (durante
2 segundos)

Cuando se pulsa

La báscula se enciende o apaga.
El indicador de espera aparece cuando se apaga la pantalla.
El modo de pesaje se activa cuando se enciende la pantalla.
Se necesita una contraseña cuando se enciende la báscula mediante la
función de contraseña.
Esta tecla está siempre disponible. Si se pulsa esta tecla durante el
funcionamiento, la pantalla se apagará.
En el modo de pesaje, la legibilidad
cambia.
La báscula entra en el modo de tabla deEn el modo de recuento o defunciones.
porcentaje, la báscula entra en elmodo de almacenamiento.
Las unidades de pesaje
Se ejecuta el modo de ajuste de la
seleccionadas en la tabla derespuesta.
funciones se cambian por orden.
28

Español (ES)TeclaCuando se mantiene pulsada (durante
2 segundos)

Cuando se pulsa

Las series HR-AZ pueden realizar elajuste de la sensibilidad utilizando
Se ejecuta el modo de ajuste de lael peso incorporado.
sensibilidad.
En el modo de tabla de funciones,
funciona como tecla de cancelación.
No tiene ninguna función en los
Los datos de pesaje estable seajustes de fábrica.
emiten según la tabla de funciones.
El “Title block (Bloque de título)” y el
En cada procedimiento de uso“End block (Bloque final)” se emiten
funciona como tecla desegún los ajustes GLP/GMP de la
determinación.
tabla de funciones.
El valor de medición actual se considera cero y se muestra el cero.

Esta función realiza el ajuste de la sensibilidad de las series HR-AZ con el peso
incorporado (la masa interna).

Procedimiento

Paso 1. Conecte el adaptador de CA a la báscula y deje que se caliente durante un
mínimo de una hora sin nada en el plato de pesaje.
Paso 2. Coloque el accesorio de protección contra corrientes de aire sobre la báscula.
Pulse la tecla CAL para mostrar.
Paso 3. El ajuste de la sensibilidad se realiza mediante la masa interna.
No permita que la báscula se vea afectada por vibraciones o corrientes de aire.
Paso 4. Si se selecciona el parámetro para utilizar la salida GLP de la tabla de funciones,
se emite un informe de ajuste de la sensibilidad después del ajuste de lasensibilidad.
Paso 5. La báscula volverá automáticamente al modo de pesaje tras el ajuste de lasensibilidad.

Español (ES)29

Esta función realiza el ajuste de la sensibilidad con una pesa externa.

Procedimiento

Paso 1. Conecte el adaptador de CA a la báscula y dejeque se caliente durante un mínimo de una hora
Mantener pulsada durante
sin nada en el plato de pesaje.
aproximadamente 2 segundos.
Paso 2. Mantenga pulsada la tecla CAL hasta que
aparezca
durante aproximadamente dos
segundos. A continuación, suelte la tecla.
Paso 3. La báscula indica

Soltar.

Si desea cambiar el peso de ajuste de lasensibilidad, pulse la tecla SAMPLE y vaya al
paso 4. Consulte “8. CALIBRATION (8.
CALIBRACIÓN)” en el manual de instrucciones
para ver los pesos de ajuste de la sensibilidad quese pueden seleccionar.
Si utiliza el valor del peso de ajuste de la sensibilidad
almacenado en la báscula, vaya al paso 5.

Ejemplo de nuevo valor

Paso 4. Especifique el valor del peso de la siguiente manera:
Tecla SAMPLE ...... Parpadean todos los segmentos: modo de selección depeso.
Parpadean los tres últimos dígitos: ajuste del valor delpeso.
Tecla RE-ZERO (+) ...Parpadean todos los segmentos: seleccionar peso.
Tecla MODE (-)
Parpadean los tres últimos dígitos: ajuste del valor delpeso.
Tecla PRINT ......... El valor del peso se almacena en la memoria. Los
datos se mantienen aunque se desconecte laalimentación.
Tecla CAL ............. Se cancela la operación y la pantalla vuelve a

Peso de ajuste dela sensibilidad.

Paso 5. Compruebe que no haya nada sobre el plato y pulse la tecla
PRINT . La báscula mide el punto del cero. No permita que labáscula se vea afectada por vibraciones o corrientes de aire. La
báscula indica el valor del peso de ajuste de la sensibilidad.
Paso 6. Coloque el peso de ajuste de la sensibilidad mostrado en elplato y pulse la tecla PRINT . La báscula mide el peso de ajustede la sensibilidad. No permita que la báscula se vea afectada por
vibraciones o corrientes de aire.
Paso 7. Retire la pesa del plato después de mostrar

Punto del cero

Salida GLP.

Paso 8. Si se ha configurado la salida GLP, se emite un “informe de calibración”.
Paso 9. La báscula volverá automáticamente al modo de pesaje.
30

Español (ES)Paso 10. Coloque el peso de ajuste de la sensibilidad en el plato y compruebe que el valor
de medición está dentro de ±2 dígitos. Si está fuera de ese intervalo, compruebe
las condiciones ambientales y repita desde el paso 2.

PantallaCódigo de error

Comprobación y soluciones

Error de estabilidad“Visualización de cero” o “ajuste de la sensibilidad” no sepueden realizar debido a un valor de pesaje inestable.
Compruebe el área alrededor del plato de pesaje. Consulte
EC,E11 “2.1. Medidas de precaución de la instalación y el pesaje”. No
permita que la báscula se vea afectada por vibraciones,
corrientes de aire, electricidad estática ni campos magnéticos.
Pulse la tecla CAL para volver al modo de pesaje.

Error de fuera del intervalo

El valor de entrada está fuera de intervalo. Introduzca un valor
dentro del intervalo.

Error de masa interna (solo para las series HR-AZ)
EC,E17

El mecanismo de la masa interna no funcionó correctamente.
Vuelva a realizar la operación.

Error del peso de ajuste de la sensibilidad(excesivo)

El peso de ajuste de la sensibilidad es demasiado alto.
EC,E20 Compruebe que el plato de pesaje está bien colocado.
Compruebe el valor del peso de ajuste de la sensibilidad. Pulse
la tecla CAL para volver al modo de pesaje.

Error del peso de ajuste de la sensibilidad(insuficiente)

El peso de ajuste de la sensibilidad es demasiado bajo.
EC,E21 Compruebe que el plato de pesaje está bien colocado.
Compruebe el valor del peso de ajuste de la sensibilidad. Pulse
la tecla CAL para volver al modo de pesaje.

Error de sobrecarga

El valor de pesaje supera la capacidad de pesaje. Retire lamuestra del plato.

Error del plato de pesaje

El valor del peso es demasiado bajo. Compruebe que el platode pesaje y el soporte del plato están bien colocados. Pulse la
tecla ON:OFF dos veces para volver al modo de pesaje. Si
persiste el error, realice el ajuste de la sensibilidad.

Error de masa de la muestra

El valor de la masa de la unidad de muestra para el modo derecuento o de porcentaje no se puede almacenar porque es
demasiado bajo. Utilice una muestra más pesada.
Español (ES)31

Pantalla

Código de error

Comprobación y soluciones

Error de masa de la unidad

El valor de la masa de la unidad de muestra para el modo derecuento es demasiado bajo.
Si el valor de la masa de la unidad se guarda tal cual, puede
provocar un error de recuento.
Añada muestras del número especificado y pulse la tecla
PRINT para obtener un pesaje preciso.

Error interno de la báscula

Si este error aparece constantemente, póngase en contacto
con su proveedor local de A&D.

Error de la batería del reloj

La batería de reserva del reloj se ha agotado por completo.
Ajuste la fecha y la hora después de pulsar una de las teclas. Si
se conecta la báscula a la fuente de alimentación, aunque labatería de reserva del reloj se haya agotado, el rendimiento dela báscula se mantendrá. Cuando este error aparece con
frecuencia, es necesario solicitar la reparación.

Error de batería baja

Ha disminuido la tensión de la fuente de alimentación.
Compruebe si se está utilizando el adaptador de CA correcto.
Si se utiliza una batería recargable, sustitúyala.

Error de comunicación

EC,E00 Ha habido un error del protocolo de comunicaciones.
Compruebe el formato, la velocidad en baudios y la paridad.

Error de comando no definido

EC,E01 Se ha recibido un comando no definido. Compruebe elcomando.

La báscula no está lista

No se puede procesar el comando recibido.
Compruebe la sincronización del comando transmitido.
EC,E02 P. ej., la báscula ha recibido un comando Q en un modo distintodel de pesaje.
P. ej., la báscula ha recibido un comando Q mientras estaba
procesando el comando RE-ZERO.

Error de tiempo límite

EC,E03Si
se ha configurado, la báscula no recibió el siguiente
carácter de un comando dentro del tiempo límite de un
segundo. Compruebe la comunicación.

Error de exceso de caracteres

EC,E04 La báscula ha recibido demasiados caracteres en un comando.
Compruebe el comando.

Error de formato

Un comando incluye datos incorrectos.
EC,E06 Compruebe el comando.
P. ej. Los datos numéricos son incorrectos.
P. ej. Los datos alfanuméricos son incorrectos.
32

Español (ES)Pantalla

Código de error

Otros errores

Comprobación y soluciones

Error del valor de ajuste

EC,E07 Los datos recibidos superan el intervalo aceptable.
Compruebe el intervalo de valores del parámetro del comando.
Si no se pueden resolver los errores descritos anteriormente oaparecen otros errores, póngase en contacto con su proveedor
local de A&D.

Capacidad de pesaje
Valor de visualización
máximoHR-100AZ
HR-100A
102 g

HR-150AZ
HR-150A
152 g

HR-250AZ
HR-250A
252 g

102,0084 g 152,0084 g

Legibilidad

252,0084 g

0,1 mgRepetibilidad(desviación estándar)

0,1 mg0,2 mg (de 200 a
250 g)
0,1 mg (de 0 a 200 g)

Linealidad±0,2 mg±0,3 mg

HR-251AZ
HR-251A
252,008 g
62,0009 g 1
1g
0,1 g
0,5 mg0,1 mg±1 mg±0,3 mg

1 La báscula permite pesar con un intervalo de precisión, incluso al colocar un valor
grande de tara en el plato. (Función de intervalo inteligente)

Español (ES)33

Questa guida introduttiva descrive le funzioni di base e le operazioni della bilancia(serie HR-AZ / HR-A). Le funzioni e le procedure operative della bilancia sono
illustrate nel dettaglio nel "HR-AZ / HR-A series Analytical Balance INSTRUCTION
MANUAL (MANUALE DI ISTRUZIONI della bilancia analitica della serie HR-AZ /HR-A)". Questo manuale può essere scaricato dal nostro sito web
https://www.aandd.jp.
2021 A&D Company, Limited. Tutti i diritti riservati.
Nessuna parte della presente pubblicazione può essere riprodotta, trasmessa,
trascritta o tradotta in qualsiasi lingua, in qualsiasi forma e con qualsiasi mezzo senza
l'autorizzazione scritta da parte di A&D Company, Limited.
Il contenuto del presente manuale e le specifiche tecniche dello strumento in esso
illustrate sono soggetti a modifiche per migliorie senza preavviso.
Gli altri marchi di fabbrica e nomi commerciali appartengono ai rispettivi proprietari.

La bilancia è uno strumento di precisione. Disimballare con
cura. Si raccomanda di conservare i materiali di imballaggio,
che andranno utilizzati per trasportare la bilancia in futuro.
Il contenuto dell'imballaggio differisce a seconda del modellodella bilancia. Vedere le illustrazioni seguenti per verificare
che sia incluso tutto.

Attenzione:

Verificare che il tipo di adattatore CA sia adatto al voltaggio locale e al tipo di presa.
Utilizzare l'apposito adattatore CA specificato per la bilancia.
Non utilizzare l'adattatore CA fornito con la bilancia per altri strumenti.
Se si utilizza un adattatore CA sbagliato, la bilancia e altri apparecchi potrebbero non
funzionare correttamente.

34

Italiano (IT)Attenzione Non installare la bilancia in presenza di gas infiammabili o corrosivi.
Per ottenere prestazioni ottimali dalla bilancia e acquisire dati di pesatura precisi,
considerare quanto segue:Installare la bilancia in un ambiente in cui la temperatura e l'umidità non siano
eccessive. La temperatura migliore per il funzionamento è di circa 20 °C ±2 °C a circa
il 45% - 60% di umidità relativa (UR).
Installare la bilancia in un luogo che sia privo di polvere.
Il tavolo di pesatura deve essere solido. (È particolarmente indicato un tavolo
antivibrazione o un tavolo di pietra)
Posizionare la bilancia su un tavolo orizzontale e accertarsi che non sia inclinata.
Installare la bilancia in un luogo stabile ed evitare vibrazioni e scosse. I luoghi migliorisono gli angoli delle stanze al primo piano, poiché sono meno soggetti a vibrazioni.
Installare la bilancia in un luogo che non sia in prossimità di termosifoni o impianti diaria condizionata.
Installare la bilancia in un luogo che non sia direttamente esposto alla luce solare.
Installare la bilancia lontano da apparecchi che producono campi magnetici.
Prima dell'uso, accertarsi di collegare l'adattatore CA alla bilancia e all'alimentazione
per almeno un'ora, affinché la bilancia si riscaldi.
Prima del primo utilizzo o dopo aver spostato la bilancia in un'altra posizione, eseguirela regolazione della sensibilità in modo da poter effettuare una pesatura precisa.
Livellare la bilancia regolando i piedini diLa bilancia è livellata.
livellamento, facendo in modo che la bolla
Pesatura corretta.
della livella a bolla d'aria sia centrata
La bilancia è inclinata.
all'interno del cerchio rosso.
Pesatura imprecisa.
SuGiùPiedino di livellamento

Livella a bolla d'aria

Livella a bolla d'aria
GiùSuPiedino di livellamento

Quando la bolla è decentrata verso sinistra Quando la bolla è decentrata verso destra
Ruotare il piedino di livellamento anteriore
Ruotare il piedino di livellamento anteriore
destro in senso orario.
sinistro in senso orario.

Quando la bolla è decentrata verso la parte Quando la bolla è decentrata verso la parteposteriore
anteriore
Ruotare contemporaneamente entrambi i
Ruotare contemporaneamente entrambi i
piedini di livellamento anteriori in senso orario. piedini di livellamento anteriori in senso
antiorario.

Italiano (IT)35

Numero di dati statistici (Modalità di calcolo statistico)
Il rapporto tra il valore di pesatura e la capacità di pesatura, in percentuale, nella
modalità di pesatura. (Indicatore di capacità)
Parametri della funzione.
Indicatore di processo

Indicatori di risposta

Indicatore di stabilizzazione

Indicatori del comparatore
Indicatore della modalità di attesa
Unità

Indicatore di standby
Valori di pesatura, dati memorizzati e vociModalità di output a intervalli.
Indicatore di standby.

Indicatore di processo
Lampeggiante

Modalità di output interno.
Indicatore attivo.

È possibile "Premere e rilasciare immediatamente il tasto" o "Tenere premuto il tasto(per 2 secondi)", il comportamento della bilancia varia a seconda di quale delle due azioni
viene eseguita.
Per la normale operazione di pesatura è necessario "Premere e rilasciare
immediatamente il tasto". Non mantenere premuto il tasto a meno che non sia
necessario.
Premere il tasto (Premere e rilasciare immediatamente il tasto) ......................
Tenere premuto il tasto (per circa 2 secondi) ..................................................
Tasto

Se mantenuto premuto(per 2 secondi)

Se premuto

È possibile accendere o spegnere la bilancia.
L'indicatore di standby viene visualizzato quando il display è spento.
La modalità di pesatura è abilitata quando il display è acceso.
È richiesta una password quando la bilancia viene accesa utilizzando lafunzione password.
Questo tasto è sempre disponibile. Se il tasto viene premuto durante
l'operazione, il display verrà spento.
In modalità di pesatura, cambia laleggibilità.
La bilancia entra in modalità tabella
In modalità di conteggio o percentuale,delle funzioni.
la bilancia entra in modalità dimemorizzazione.
36

Italiano (IT)TastoSe mantenuto premuto(per 2 secondi)

Se premuto

Le unità di pesatura selezionate nella
Viene eseguita la modalità ditabella delle funzioni vengono
regolazione della risposta.
commutate in ordine.
La serie HR-AZ può eseguire la
regolazione della sensibilità utilizzando
Viene eseguita la modalità diil peso incorporato.
regolazione della sensibilità.
In modalità tabella delle funzioni,
funziona come tasto di annullamento.
Nessuna funzione per
Vengono inviati i dati di pesatura
impostazione di fabbrica.
stabile in base alla tabella delle
Il "Title block (Blocco del titolo)" e ilfunzioni.
"End block (Blocco finale)"
In ogni procedura operativa, funziona
vengono inviati in base alle
come tasto di determinazione.
impostazioni GLP/GMP nella
tabella delle funzioni.
Il valore di misurazione corrente è considerato zero e viene visualizzato zero.

Questa funzione esegue la regolazione della sensibilità della serie HR-AZ utilizzando ilpeso incorporato (la massa interna).

Procedura

Passaggio 1. Collegare l'adattatore CA alla bilancia e far riscaldare la bilancia peralmeno un'ora senza posizionare niente sul piatto per la pesatura.
Passaggio 2. Posizionare il frangivento accessorio sulla bilancia.
Premere il tasto CAL per visualizzare.
Passaggio 3. La regolazione della sensibilità viene eseguita utilizzando la massa interna.
Non esporre a vibrazioni o correnti d'aria che influirebbero sulla bilancia.
Passaggio 4. Se si sceglie il parametro per utilizzare l'output GLP della tabella delle
funzioni, al termine della regolazione della sensibilità verrà inviato un
rapporto di regolazione della sensibilità.
Passaggio 5. Dopo la regolazione della sensibilità, la bilancia ritornerà automaticamente
alla modalità di pesatura.

Italiano (IT)37

Questa funzione esegue la regolazione della sensibilità utilizzando un peso esterno.

Procedura

Passaggio 1. Collegare l'adattatore CA alla bilancia e farla
riscaldare per almeno un'ora senza posizionare
niente sul piatto per la pesatura.

Tenere premuto per
circa 2 secondi.

Passaggio 2. Tenere premuto il tasto CAL per circa due
secondi finché non viene visualizzato. Quindi rilasciare il tasto.
Passaggio 3. La bilancia visualizza

Rilasciare.

Se si desidera cambiare il peso diregolazione della sensibilità, premere il tasto
SAMPLE e procedere al passaggio 4.
Consultare "8. CALIBRATION (8.
CALIBRAZIONE)" nel manuale di istruzioni
per i pesi di regolazione della sensibilità cheè possibile selezionare.

Esempio nuovo valore

Se si utilizza il valore del peso di regolazione dellasensibilità memorizzato nella bilancia, procedere alpassaggio 5.

Punto zeroPassaggio 4. Specificare il valore del peso come segue:
Tasto SAMPLE .... Tutti i segmenti lampeggiano : modalità di selezione delpeso.
Le ultime tre cifre lampeggiano : regolazione del valoredel peso.
Tasto RE-ZERO (+) Tutti i segmenti lampeggiano : selezionare il peso.
Tasto MODE (-) Le ultime tre cifre lampeggiano : regolazione del valoredel peso.

Peso diregolazionedella sensibilità

Tasto PRINT ....... Il valore del peso è memorizzato in memoria. I dati
vengono mantenuti anche se si stacca l'alimentazione.
Tasto CAL ........... L'operazione viene annullata e il display ritorna a.

Passaggio 5. Verificare che non ci sia niente sul piatto e premere iltasto PRINT. . La bilancia misura il punto zero. Nonesporre a vibrazioni o correnti d'aria che influirebbero
sulla bilancia. La bilancia visualizza il valore del peso diregolazione della sensibilità.

Output GLP

Passaggio 6. Posizionare il peso di regolazione della sensibilità
visualizzato sul piatto e premere il tasto PRINT . La
bilancia misura il peso di regolazione della sensibilità. Nonesporre a vibrazioni o correnti d'aria che influirebbero sullabilancia.
Passaggio 7. Rimuovere il peso dal piatto dopo aver visualizzato
38.

Italiano (IT)Passaggio 8. Se l'output GLP è impostato, viene inviato il "Rapporto di calibrazione".
Passaggio 9. La bilancia ritornerà automaticamente alla modalità di pesatura.
Passaggio 10. Posizionare il peso di regolazione della sensibilità sul piatto e verificare che
il valore di misurazione sia entro ±2 cifre. Se è al di fuori di questo
intervallo, controllare le condizioni ambientali e ripetere la procedura dalpassaggio 2.

Display

Codice di errore

Controllo e soluzioni

Errore di stabilità

Non è possibile eseguire la "visualizzazione zero" o la"regolazione della sensibilità" a causa di un valore di pesatura
instabile. Controllare l'area intorno al piatto per la pesatura.
EC,E11 Consultare "2.1. Precauzioni per l'installazione e la pesatura".
Evitare che vibrazioni, correnti d'aria, elettricità statica e campi
magnetici influiscano sulla bilancia.
Premere il tasto CAL per ritornare alla modalità di pesatura.

Errore fuori intervallo

Il valore immesso è al di fuori dell'intervallo. Immettere un valore
che rientri nell'intervallo.

Errore della massa interna (solo per la serie HRAZ)

EC,E17 Il meccanismo della massa interna non ha funzionato
correttamente. Eseguire nuovamente l'operazione.

Errore del peso di regolazione della sensibilità(valore positivo)

Il peso di regolazione della sensibilità è troppo pesante.
Verificare che il piatto per la pesatura sia installato
EC,E20correttamente. Verificare il valore del peso di regolazione dellasensibilità. Premere il tasto CAL per ritornare alla modalità dipesatura.

Errore del peso di regolazione della sensibilità(valore negativo)

Il peso di regolazione della sensibilità è troppo leggero.
Verificare che il piatto per la pesatura sia installato
EC,E21
correttamente. Verificare il valore del peso di regolazione dellasensibilità. Premere il tasto CAL per ritornare alla modalità dipesatura.

Errore di sovraccarico

Il valore di pesatura supera la capacità di pesatura. Rimuovere
il campione dal piatto.

Italiano (IT)39

Display

Codice di errore

Controllo e soluzioni

Errore del piatto per la pesatura

Il valore del peso è troppo basso. Verificare che il piatto per lapesatura e il supporto del piatto siano installati correttamente.
Premere due volte il tasto ON:OFF per ritornare alla modalità dipesatura. Se l'errore persiste, eseguire la regolazione dellasensibilità.

Errore della massa del campioneNon è possibile memorizzare il valore della massa dell'unità dicampionamento per la modalità di conteggio o per la modalità
percentuale perché è troppo basso. Utilizzare un campione piùpesante.

Errore della massa dell'unità

Il valore della massa dell'unità di campionamento per lamodalità di conteggio è troppo basso.
Se il valore della massa dell'unità viene salvato così com'è,
possono verificarsi errori di conteggio.
Aggiungere il numero specificato di campioni e premere il tasto
PRINT per acquisire una pesatura precisa.

Errore interno della bilancia

Se questo errore viene visualizzato i modo persistente,contattare il rivenditore locale di A&D.

Errore della batteria dell'orologio

La batteria di riserva dell'orologio si è scaricata completamente.
Regolare la data e l'ora dopo aver premuto uno dei tasti. Se sicollega la bilancia all'alimentazione, anche se la batteria diriserva dell'orologio si è scaricata, le prestazioni della bilancia
verranno mantenute. Quando questo errore viene visualizzato
di frequente, è necessaria la riparazione.

Errore di batteria scarica

La tensione di alimentazione è diminuita.
Verificare che venga utilizzato l'adattatore CA corretto.
Se si utilizza una batteria ricaricabile, sostituirla.

Errore nelle comunicazioni

EC,E00 Si è verificato un errore di protocollo nelle comunicazioni.
Controllare il formato, la velocità in baud e la parità.

Errore di comando non definito

EC,E01 È stato ricevuto un comando non definito. Controllare ilcomando.

Non pronto

Un comando ricevuto non può essere elaborato.
Controllare la tempistica del comando trasmesso.
EC,E02 Es. La bilancia ha ricevuto un comando Q in una modalità
diversa da quella di pesatura.
Es. La bilancia ha ricevuto un comando Q mentre stava
elaborando un comando RE-ZERO.
40

Italiano (IT)Display

Codice di errore
EC,E03

Errore di timeout

Controllo e soluzioni

Quando è impostato, la bilancia non ha ricevuto ilcarattere successivo di un comando entro il tempo limite di un
secondo. Verificare la comunicazione.

Errore di eccesso di caratteri
EC,E04

La bilancia ha ricevuto un numero eccessivo di caratteri in un
comando.
Controllare il comando.

Errore di formato

Un comando include dati non corretti.
EC,E06 Controllare il comando.
Es. I dati numerici sono errati.
Es. I dati alfanumerici sono errati.

Errore nell'impostazione del parametro

Altri errori

EC,E07 I dati ricevuti superano l'intervallo accettabile.
Verificare l'intervallo del parametro del comando.
Se non è possibile risolvere gli errori descritti in precedenza ose vengono visualizzati altri errori, contattare il rivenditore locale
di A&D.

Capacità di pesatura

HR-100AZ
HR-100A
102 g

HR-150AZ
HR-150A
152 g

Display massimo

102,0084 g 152,0084 g

0,1 mgLinearità±0,2 mg

HR-251AZ
HR-251A

252,008 g
62,0009 g 1
1g
0,1 mg0,1 g
0,5 mg0,2 mg (da 200 a 250 g)
0,1 mg (da 0 a 200 g)
0,1 mg±1 mg±0,3 mg±0,3 mg

Leggibilità
Ripetibilità(Deviazione standard)

HR-250AZ
HR-250A
252 g
252,0084 g

1 La bilancia consente di eseguire la pesatura utilizzando un intervallo di precisione,
anche con una tara pesante posizionata sul piatto. (Funzione di intervallo Smart)

Italiano (IT)41

此快速入門指南說明天平（HR-AZ／HR-A 系列）的基本功能和操作方式。「HR-AZ /HR-A series Analytical Balance INSTRUCTION MANUAL（HR-AZ／HR-A 系列分
析天平說明手冊）」中詳細說明天平的功能和操作程序。您可以從我們的網站下載此手
冊： https://www.aandd.jp。
2021 A&D Company, Limited。保留所有權利。
未經 A&D Company, Limited 書面許可，不得透過任何方式以任何形式複製、傳播、轉
錄本出版物的任何部分或翻譯成任何語言。
本手冊的內容以及本手冊涵蓋的儀器規格可能因改進而隨時變更，恕不另行通知。
其他商標和商號歸其各自所有者所有。

天平是精密儀器。拆開包裝時請小心，建議保留包裝材料，以便將來搬運天平時使用。
包裝內容物因天平型號而異。請參閱以下插圖，確認沒有任何
遺漏。

警告：

請確認 AC 變壓器類型適合當地電壓和插座類型。
請使用天平指定的專用 AC 變壓器。
請勿將天平隨附的 AC 變壓器用於其他儀器。
如果使用錯誤的 AC 變壓器，天平及其他設備可能無法正常運作。

42

中文(CT)

警告

請勿將天平安裝在有可燃性或腐蝕性氣體的位置。

為了讓天平充分發揮效能並產生準確的稱重資料，請注意以下事項：
將天平安裝在溫濕度適中的環境。最佳工作溫度約為 20 °C ±2 °C，相對濕度約為 45%
至 60% RH。
將天平安裝在無灰塵的位置。
稱量台應為堅固的固體（最理想的是抗震的桌子或石桌）。
將天平放在水平桌上，並確保未傾斜。
將天平安裝在平穩的位置，避免震動和撞擊。最好放在一樓房間的角落，比較不會受到
震動。
將天平安裝在不受暖氣或空調影響的位置。
將天平安裝在不受陽光直射的位置。
切勿將天平安裝在會產生磁場的設備附近。
請務必在使用前將 AC 變壓器連接至天平和電源至少一小時，讓天平暖機。
初次使用前或移動至其他位置後請執行靈敏度調整，以進行準確的稱重。
調整水平調整腳座以讓天平呈水平，使氣泡天平呈水平。
稱重正確。
式水平儀的氣泡置中於紅色圓圈內。
天平傾斜。
稱重不準確。
氣泡式水平儀上下水平調整腳座氣泡式水平儀下上水平調整腳座氣泡向左偏移時
將右前方的水平調整腳座依順時針方向旋
轉。

氣泡向右偏移時
將左前方的水平調整腳座依順時針方向旋轉。

氣泡向後偏移時
同時將前方的兩個水平調整腳座依順時針方向旋轉。

氣泡向前偏移時
同時將前方的兩個水平調整腳座依逆時針方向
旋轉。

中文(CT)

43

統計資料數量（統計計算模式）
稱重模式下稱重值與稱重上限之間的比例，以百分比表示。（上限指示燈）
功能的參數。
處理指示燈

反應指示燈

穩定性指示燈比較器指示燈
保持模式指示燈
單位

待機指示燈
稱重值、已儲存資料和項目

間隔輸出模式。
待機指示燈。

處理指示燈
閃爍內部輸出模式。
使用中指示燈。

天平的操作模式依據「按下並立即放開按鍵」或「長按按鍵（2 秒）」的行為而不同。
一般稱重操作是「按下並立即放開按鍵」。除非需要時才按住按鍵。
按下按鍵（按下並立即放開按鍵。） .................................................................
長按按鍵（大約 2 秒） ......................................................................................
按鍵

按下時
長按（2 秒）
能夠開啟或關閉天平。
關閉顯示幕時會顯示待機指示燈。
開啟顯示幕時會啟用稱重模式。
使用密碼功能開啟天平時，必須輸入密碼。
可隨時使用此按鍵。如果在操作過程中按此按鍵，將關閉顯示器。
在稱重模式下，可讀性會改變。
在計數或百分比模式下，天平會進入 天平會進入功能表模式。
儲存模式。
依序切換在功能表上選擇的稱重單
執行反應調整模式。
位。
HR-AZ 系列可使用內建砝碼執行靈敏度調整。
執行靈敏度調整模式。
在功能表模式下，作為取消鍵。
在出廠設定下無功能。
根據功能表輸出穩定的稱重資料。
根據功能表上的 GLP/GMP 設定輸出
在各個操作程序中，作為確定鍵。
「Title block（標題區塊）」和「Endblock（結束區塊）」。
將目前測量值視為零並顯示零。
44

中文(CT)

此功能使用內建砝碼（內部質量）執行 HR-AZ 系列靈敏度調整。

程序步驟 1. 將 AC 變壓器連接至天平，並讓天平暖機至少一小時，長按勿將任何物體放在秤盤上。
步驟 2. 將配件擋風板放在天平上。
按下 CAL 按鍵顯示。

步驟 3. 使用內部質量執行靈敏度調整。
請勿讓震動或氣流影響天平。
步驟 4. 如果選擇參數以使用功能表的 GLP 輸出，則靈敏度調整之後，會輸出靈敏度調整報
告。
步驟 5. 靈敏度調整之後，天平將自動返回稱重模式。

此功能使用外部砝碼執行靈敏度調整。

程序步驟 1. 將 AC 變壓器連接至天平，並暖機至少一小時，此時勿將任何物體放在秤盤上。
按住大約 2 秒
步驟 2. 長按住 CAL 按鍵大約兩秒，直到顯示
後放開按鍵。
步驟 3. 天平顯示

。然
放開

。

若要更換靈敏度調整砝碼，按下 SAMPLE 按鍵並進
入步驟 4。請參閱說明手冊的「8. CALIBRATION（8. 校正）」瞭解可選的靈敏度調整砝碼。
如果使用儲存在天平中的靈敏度調整砝碼值，請進入
步驟 5。
步驟 4. 如下所述指定砝碼值：

新值例子
零點

SAMPLE 按鍵............所有區段閃爍：砝碼選擇模式。

最後三個數字閃爍：砝碼值調整。
RE-ZERO 按鍵 (+) ....所有區段閃爍：選擇砝碼。
MODE 按鍵 (-)
最後三個數字閃爍：砝碼值調整。
PRINT 按鍵 ................砝碼值儲存在記憶體中。即使切斷電

源，也會保存資料。
CAL 按鍵....................取消操作，顯示器返回
中文(CT)。
45

靈敏度調整砝碼

步驟 5. 確認秤盤上無任何物體，按下 PRINT 按鍵。天平測量零點。
請勿讓震動或氣流影響天平。天平顯示靈敏度調整砝碼值。
步驟 6. 將顯示的靈敏度調整砝碼放在秤盤上，按下 PRINT 按鍵。
天平測量靈敏度調整砝碼。請勿讓震動或氣流影響天平。
步驟 7. 顯示之後，從秤盤上取下砝碼。

步驟 8. 若已設定 GLP 輸出，則會輸出「校正報告」。
步驟 9. 天平將自動返回稱重模式。

GLP 輸出步驟 10. 將靈敏度調整砝碼放在秤盤上，確認測量值在 ±2 位數範圍內。
如果超出範圍，則檢查環境條件並從步驟 2 開始重複。

顯示

錯誤代碼

檢查和解決方法

穩定性錯誤
因稱重值不穩定而無法執行「零顯示」或「靈敏度調整」。檢查
秤盤周圍的區域。請參閱「2.1.安裝和稱重的注意事項」。防止震EC,E11動、氣流、靜電和磁場影響天平。
按下 CAL 按鍵返回稱重模式。

超出範圍錯誤輸入值超出範圍。輸入範圍內的值。

內部質量錯誤（僅適用於 HR-AZ 系列）

EC,E17 內部質量的機制無法正常運作。再次執行操作。

靈敏度調整砝碼錯誤（正值）靈敏度調整砝碼過重。
EC,E20 確認已正確安裝秤盤。確認靈敏度調整砝碼值。按下 CAL 按鍵
返回稱重模式。

靈敏度調整砝碼錯誤（負值）靈敏度調整砝碼過輕。
EC,E21 確認已正確安裝秤盤。確認靈敏度調整砝碼值。按下 CAL 按鍵
返回稱重模式。

超載錯誤
稱重值超過稱重上限。從秤盤取下樣品。

秤盤錯誤
重量值過輕。確認已正確安裝秤盤和秤盤支架。按下 ON:OFF按鍵兩次以返回稱重模式。如果錯誤仍然存在，請執行靈敏度調整。

樣品質量錯誤
無法儲存計數模式或百分比模式的樣品單位質量值，因為過輕。
請使用較重的樣品。

46

中文(CT)

顯示

錯誤代碼

檢查和解決方法

單位質量錯誤計數模式的樣品單位質量值過輕。
如果照樣儲存單位質量值，可能會導致計數錯誤。
增加指定數量的樣品，按下 PRINT 按鍵取得準確的稱重。

天平內部錯誤

如果此錯誤持續出現，請聯絡當地 A&D 經銷商。

時鐘電池錯誤時鐘的備用電池電量已完全耗盡。按下任意按鍵後，調整日期和時間。如果將天平連接至電源，即使時鐘的備用電池電量耗盡，仍會維持天平性能。此錯誤頻繁出現時，需要維修。

電量不足錯誤電源電壓下降。
確認是否使用正確的 AC 變壓器。
如果使用充電式電池，請更換。

通訊錯誤

EC,E00 通訊時發生通訊協定錯誤。
請確認格式、鮑率和奇偶。
EC,E01

未定義的命令錯誤
收到未定義的命令。請確認命令。

尚未準備就緒無法處理收到的命令。
EC,E02 檢查傳輸命令的時機。
例如 天平在稱重模式以外的模式下收到 Q 命令。
例如 天平在處理 RE-ZERO 命令時收到 Q 命令。

逾時錯誤EC,E03 設定後，天平未在一秒的時限內收到命令的下一個字元。請確認通訊情況。

超出字元錯誤EC,E04 天平收到的命令中含有過多字元。
請確認命令。

格式錯誤
命令包含不正確的資料。
EC,E06 請確認命令。
例如 數值資料不正確。
例如 英數字元資料不正確。

參數設定錯誤其他錯誤中文(CT)

EC,E07 收到的資料超出可接受範圍。
請確認命令的參數範圍。
如果無法解決上述錯誤或顯示其他錯誤，請聯絡當地 A&D 經銷
商。

47

稱重上限

HR-100AZ
HR-100A
102 g

HR-150AZ
HR-150A
152 g

最大顯示數值

102.0084 g 152.0084 g

0.1 mg線性±0.2 mg

HR-251AZ
HR-251A

252.008 g
62.0009 g 1
1g
0.1 mg0.1 g
0.5 mg
0.2 mg（200 至 250 g）
0.1 mg（0 至 200 g）
0.1 mg±1 mg±0.3 mg±0.3 mg

可讀性
可重複性
（標準差）

HR-250AZ
HR-250A
252 g
252.0084 g

1 天平允許使用精密範圍進行稱重，即使將較重的皮重放在秤盤上也不成問題。（智慧範
圍功能）

48

中文(CT)

MEMO

49

MEMO

50

MEMO

51

Back cover: A&D global

A&D Company, Limited
3-23-14 Higashi-Ikebukuro, Toshima-ku, Tokyo 170-0013, JAPAN
Telephone: [81] (3) 5391-6132
Fax: [81] (3) 5391-1566
A&D ENGINEERING, INC.
1756 Automation Parkway, San Jose, California 95131, U.S.A.
Telephone: [1] (408) 263-5333
Fax: [1] (408)263-0119
A&D INSTRUMENTS LIMITED
Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire OX14 1DY
United Kingdom
Telephone: [44] (1235) 550420
Fax: [44] (1235) 550485
A&D AUSTRALASIA PTY LTD
32 Dew Street, Thebarton, South Australia 5031, AUSTRALIA
Telephone: [61] (8) 8301-8100
Fax: [61] (8) 8352-7409
A&D KOREA Limited
한국에이.엔.디(주)서울특별시 영등포구 국제금융로6길33 (여의도동) 맨하탄빌딩 817 우편 번호 07331( 817, Manhattan Bldg., 33. Gukjegeumyung-ro 6-gil, Yeongdeungpo-gu, Seoul, 07331
Korea )
전화: [82] (2) 780-4101
팩스: [82] (2) 782-4264
OOO A&D RUS
OOO "ЭЙ энд ДИ РУС"Почтовый адрес:121357, Российская Федерация, г.Москва, ул. Верейская, дом 17Юридический адрес: 117545, Российская Федерация, г. Москва, ул. Дорожная, д.3,
корп.6, комн. 8б
( 121357, Russian Federation, Moscow, Vereyskaya Street 17 )тел.: [7] (495) 937-33-44
факс: [7] (495) 937-55-66
A&D Instruments India Private Limited( 509, Udyog Vihar, Phase- , Gurgaon - 122016, Haryana, India ): [91] (124) 4715555: [91] (124) 4715599
A&D SCIENTECH TAIWAN LIMITED. A&D台灣分公司 艾安得股份有限公司
台湾台北市中正區青島東路 5 號 4 樓
( 4F No.5 Ching Tao East Road, Taipei Taiwan R.O.C. )Tel : [886](02) 2322-4722
Fax : [886](02) 2392-1794
A&D INSTRUMENTS (THAILAND) LIMITEDบริษัท เอ แอนด ดี อินสทรูเมนท (ไทยแลนด) จํากัด
168/16 หมูที่ 1 ตําบลรังสิต อําเภอธัญบุรี จังหวัดปทุมธานี 12110 ประเทศไทย
( 168/16 Moo 1, Rangsit, Thanyaburi, Pathumthani 12110 Thailand )Tel : [66] 20038911
52