Multi-Functional Platform Scales

HV/W-C& HV/W-CPSeries

HV-C/HW-C series

HV-CP/HW-CP series

For The Maximum
Possible Throughput
ISO9000 CERTIFIED...Clearly a Better Value
http://www.aandd.jp

To Factory Managers Who Wish To Reduce Loss And
Increase Productivity But Don't Know Where To Start
Simply changing scales can make dramatic differences. Is that really true? Here's how...

High-speed measurement—stabilization within 1 second (for theHV-C/CP series) *1
Fast-paced weighing is an essential prerequisite for streamlining operations.

*1 With the HW-C/CP series, stabilization takes not more than 1.5 seconds.

Bright and multi-colored 5-step LED comparator lights
With the clearly-discernible LED signals, check weighing (judging whether the measured
weight is within the designated limits) becomes far more accurate and efficient than anumeric display alone, leading to a significant cost reduction. Comparison can be made inthree levels, five levels, or even seven levels* 2 for both normal weighing and piece counting.
■ Auto-tare function

LED comparator lights

・・・・・・・・・・・・・LCD display
with white backlight・・・・・The scale can perform tare*3 automatically when theweight reaches the set target range, saving the operator
from the hassle of pressing the TARE key each time forconsecutive check weighing such as filling multiple cups.

*2 With no light indicating either end of the continuum*3 To offset the weight of the object on the scale and return the display to zero

・・・・・・・

Negative values are also allowed for comparator limits.
When combined with the auto-tare function, thisenables "take-away" check weighing; that is, measuring
out the target amount repeatedly by taking it out of its
container with no key press.

Alphanumerickeypad

・・・・・・・・・・・・・・・・・・・・・■ Negative comparison

Built-in printer(for the HV/W-CP series)

Highly-visible, LCD display (25 mm character height) with white backlight
Do not underestimate the effect of display visibility on operators' work efficiency, fatigue
and tendency to make errors, especially in areas where illumination is not sufficient.

Quick and easy settings using the alphanumeric keypad
The direct alphanumeric input makes it simple and straightforward to enter/re-enter preset
tare values, comparator limits, calibration values and various internal settings.
■ Key lock function
You can inhibit the use of certain keys to prevent operators from making unwanted changes tothe scale settings.

Optional roller conveyers
Large and medium-size roller conveyers (HVW-13 andHVW-14 respectively) can make measurements ofheavy objects in production lines smoother andquicker.
HVW-13

Built-in printer for creating paper-based records on the spot (for theHV/W-CP series)
In addition to normal printing, the printer enables GMP compliant output (including date& time* 4, and the ID number of six alphanumeric characters that you set for the scale) andlabel printing without connecting to a separate printer.* 5
the HV/W-C series (version with no built-in printer) doesn't have a clock & calendar*4 Although
function, date and time can be printed using the optional AD-8127 compact printer.
printing is not supported. It becomes possible, however, by using the Universal Flexi Coms
*5 Barcode(UFC) function, which lets you edit serial data output/printout format with an external printer.

Wide selection of capacities and resolutions
Select a scale that best suits your application and budget from either HV-C/CP series of three multi-interval* 6models and HW-C/CP series of four high-resolution models (see the specifications for details).

*6 The lighter the target object is, the finer the readability becomes.
IP65 dust & waterproof base unit with a stainlesssteel (SUS430) pan
The base unit can be washed with water, not to mention that wetobjects can be weighed with no problem. Further, the surface of thepan is resistant to chemicals, scratches and rust, and easy to keep
clean and hygienic.

Three slots for optional data communicationinterfaces

IP65 base unit with a SUS430 pan

Output data, enter commands, and change the scale settings
externally via the USB (HVW-02CB)* 7 or RS-232C (HVW-03C)
interface. Either two USB interfaces, two RS-232C interfaces, or one ofeach can be installed, in addition to a comparator relay
output/buzzer/contact input interface (HVW-04C), which enables
signaling check weighing results by buzzer and/or externalcomparator (traffic) lights, as well as external key input (up to twocommands).
communication with a PC only. The driver*7 Forsoftware can be downloaded for free.

Display unit with HVW-02CB,
HVW-03C, and HVW-04C

Other Useful Features✔ Display hold function that makes it easy to read and record the display value when
external disturbances can occur unexpectedly✔ Counting mode to count a number of small pieces of the same mass value instantly andprecisely✔ Accumulation (M+) function to display the sum total of separate weighing results✔ Operation with either AC adapter or four D batteries (for the HV/W-C series)* 8✔ Display angle adjustable to four levels to suit the convenience of individual operators✔ Optional 5 m extension cable (HVW-08C) and wall-mount bracket (HVW-11C) for thedisplay unit available✔ Display cover that protects the display unit from dirt and splashes provided as standard
battery life is approx. 1200 hours using alkaline batteries, with display backlight and*8 TheLED comparator lights OFF. The HV/W-CP series operates on AC adapter only.

Specificationskglbozlb-oz

HV-15KC HV-60KC HV-200KC HW-10KC HW-60KC HW-100KC HW-200KC
HV-15KCP HV-60KCP HV-200KCP HW-10KCP HW-60KCP HW-100KCP HW-200KCPCapacity

3 / 6 / 15*i15 / 30 / 60*i 60 / 150 / 220*iMin. display

0.001 / 0.002 / 0.005* 0.005 / 0.01 / 0.02* 0.02 / 0.05 / 0.1*

Capacity

6 / 15 / 30*20150200500

Min. display

0.002 / 0.005 / 0.01*i 0.01 / 0.02 / 0.05*i 0.05 / 0.1 / 0.2*i0.0020.010.020.05

Capacity

96 / 240 / 480*i 480 / 960 / 2400*i 2400 / 4800 / 8000*i

320

2400

3200

8000

Min. display

0.05 / 0.1 / 0.2* 0.2 / 0.5 / 1*

0.05

0.2

0.5

1i

i

30 / 60 / 150* 150 / 300 / 500*iii

Capacity

30 lb

Min. display

0.1 ozi

1 / 2 / 5*ii120,000

1060

100

220

0.001

0.005

0.01

0.02

20 lb

N/A

150,000

Counting mode Max. countable numberi

N/A

0.1 oz100,000

110,000

120,000

100,000

110,000

5, 10, 20, 50 or 100

Number of samples
Stabilization time* ii

Within 1.5 seconds

Within 1 second

Repeatability (std. deviation) [kg]0.001 / 0.002 / 0.005 0.005 / 0.01 / 0.02 0.02 / 0.05 / 0.1

0.002

0.010.02

0.04

Linearity [kg]±0.001 / ±0.002 / ±0.005 ±0.005 / ±0.01 / ±0.02 ±0.02 / ±0.05 / ±0.1

±0.002

±0.01

±0.02

±0.04

±20 ppm / ˚C (5 ˚C to 35 ˚C / 41 ˚F to 95 ˚F)Sensitivity drift
Display

7 segment LCD (Character height: 25 mm) with backlight
3-color, 5-step LED comparator lights
10 times / second

Display update–10 ˚C to 40 ˚C / 14 ˚F to 104 ˚F, 85% R.H. or less (no condensation)

Operating temperaturePower supply

HV/W-C series

AC adapter or four D batteries (approx. 1200 hours using alkaline batteries, with display backlight and LED comparator lights OFF)

HV/W-CP series

AC adapter

Weighing pan size
Dimensions (W × D × H)

250 × 250 mm 330 × 424 mm 390 × 530 mm 250 × 250 mm 330 × 424 mm

390 × 530 mm

9.8 × 9.8 in 13.0 × 16.7 in 15.4 × 20.9 in 9.8 × 9.8 in 13.0 × 16.7 in

15.4 × 20.9 in

275 × 474 × 368 mm 330 × 621 × 771 mm 390 × 712 × 773 mm 275 × 474 × 368 mm 330 × 621 × 771 mm

390 × 712 × 773 mm

10.8 × 18.7 × 14.5 in 13.0 × 24.5 × 30.4 in 15.4 × 28.0 × 30.4 in 10.8 × 18.7 × 14.5 in 13.0 × 24.5 × 30.4 in

15.4 × 28.0 × 30.4 in

Weight (approx.) HV/W-C series

6 kg / 13 lb

11 kg / 24 lb 17 kg / 37 lb

6 kg / 13 lb

11 kg / 24 lb

17 kg / 37 lb

HV/W-CP series

7 kg / 15 lb

12 kg / 26 lb 18 kg / 40 lb

7 kg / 15 lb

12 kg / 26 lb

18 kg / 40 lb** to the weighing environment, response characteristic setting and how the target object is placed.

i Multi-interval: The lighter the target object is, the finer the readability becomes.
ii Time required for the last digit to settle (the stabilization mark will be lit one second after), which is subject

Options
HVW-02CB
HVW-03C
HVW-04C
HVW-08C
HVW-11C
HVW-13
HVW-14

Accessories
USB interface (bi-directional, type-B)
RS-232C interface (D-Sub 9P)
Comparator relay output/buzzer/contact input interface
Extension cable (5 m)*iii
Wall-mount bracket
Large roller conveyer (for the 100 to 200 kg capacity models)Medium-size roller conveyer (for the 60 kg capacity models)

AD-8127
AD-8526
AD-8527
AD-8920A
AD-8922A
AX-PP147-S
AX-SW128

Compact printerSerial/Ethernet converter
Quick USB adapter
Remote display
Remote controller
Paper for the built-in printer (5 rolls)
Foot switch*iii Recalibration is required after connecting the extension cable.

HVW-02CB

HVW-03C

HVW-04C
Optional

129ON

36

Clock & Calendar

HV/W-CPCounting
Function

Comparator
Function

Auto PowerON

HV/W-C

Universal
Flexi Coms

IP65

Platform
onlyUSB
Interface...Clearly a Better Value
3-23-14 Higashi-lkebukuro, Toshima-ku,Tokyo 170-0013 JAPAN
Telephone:[81](3) 5391-6132 Fax:[81](3) 5391-1566
http://www.aandd.jp
1756 Automation Parkway, San Jose, CA 95131 U.S.A.
Telephone:[1](408) 263-5333 Fax:[1](408) 263-0119

A&D Australasia Pty Ltd.

32 Dew Street, Thebarton, South Australia 5031 AUSTRALIA
Telephone:[61](8) 8301-8100 Fax:[61](8) 8352-7409

Unit 24/26 Blacklands Way Abingdon Business Park,Abingdon, Oxon OX14 1DY UNITED KINGDOM
Telephone:[44](1235) 550420 Fax:[44](1235) 550485<German Sales Office>Hamburger Straße 30 D-22926 Ahrensburg GERMANY
Telephone:[49](0) 4102 459230 Fax:[49](0) 4102 459231

Vereyskaya str.17, Moscow, 121357 RUSSIA
Telephone: [7] (495) 937-33-44 Fax: [7] (495) 937-55-66

A&D Instruments India Private Limited

509 Udyog Vihar Phase V
Gurgaon-122 016, Haryana, INDIA
Telephone: [91](124) 471-5555 Fax: [91](124) 471-5599

817, Manhattan Bldg., 33, Gukjegeumyung-ro 6-gil,Yeongdeungpo-gu, Seoul, 07331, KOREA
Telephone:[82](2) 780-4101 Fax:[82](2) 782-4280* HVW-C Series-ADCC-02-PR2-18501GP