Portable Bench Scales

SA■ One scale for both precise (0.001 kg/1 g readability*i )and high-capacity weighing■ Broad utility for home, office, floor or out in the field* i Up to 3 kg/3000 g (smart range)

Features✔ Easy to carry around and/or take aboard a vehicle to use at various locations✔ Low profile (approx. 73 mm) that can be stored in narrow spaces or propped against a wall
✔ Four weighing ranges (readabilities) selectable depending on your required precision✔ LCD display with backlight (24 mm character height), which is 180-degree reversible✔ Display hold, buzzer, auto power ON/OFF functions✔ Automatic zero setting mode that automatically
performs tare and sets the display to zero once theweight value becomes stable * ii

◀◀
◀◀

◀◀

◀◀

◀◀

◀◀

◀

◀

✔ Operation with either AC adapter (provided as standard)or four D batteries✔ Data output to a PC, printer, etc., via RS-232C interface(optional)
180-degree reversible display

This is useful, for example, when taring your own body weight before* ii measuring
the weight of an animal by holding it (you don't have to crouch
down to press the Re-Zero/AT key).

ONLCD
Backlight

Auto PowerOFFAuto PowerONSpecifications
Weighing capacity

Smart Range

SoftwareSA-30K

SA-60K

SA-150K

30 kg

60 kg

150 kgMinimum weighing value Range 1(selectable)
Range 2

Optional

RS-232C
Interface

Options
SA-03 RS-232C interface

0.1 kg0.01 kg

Dimensions (mm/inches)

Range 30.001 kg (0 to 3 kg) / 0.01 kg (greater than 3 kg)*

Range 4

1 g (0 to 3000 g) / 0.01 kg (greater than 3 kg)*

330/12.99

73/2.87~
84/3.312g

Repeatability (standard deviation)±20 g±10 g

Linearity±50 g±20 ppm / ˚C (5 ˚C to 35 ˚C / 41 ˚F to 95 ˚F)Sensitivity drift
Display

NormalBattery

7 segment LCD (character height: 24 mm) with white backlight
180˚ reversible display
Approx. 10 times / second

Display update

508.9/20.04
330/12.99

External
Calibration

Operating temperature and humidity –10 ˚C to 40 ˚C / 14 ˚F to 104 ˚F, 85% RH or less (no condensation)
Power supply

AC adapter or 4 D batteries

Battery life

Approx. 700 hours (with alkaline batteries, backlight off)
330 x 330 mm / 12.99 x 12.99 in

Weighing pan size

Approx. 6.3 kg / 13.89 lb

WeightInstruction manual, AC adapter

Standard accessories

Smart range function: The minimum weighing value will switch to 0.01 kg automatically when the display value* exceeds
3 kg/3000 g but returns to 0.001 kg/1 g by pressing the Re-Zero/AT (tare) key.

...Clearly a Better Value
3-23-14 Higashi-lkebukuro, Toshima-ku,Tokyo 170-0013 JAPAN
Telephone:[81](3) 5391-6132 Fax:[81](3) 5391-6148
http://www.aandd.jp

Unit 24/26 Blacklands Way Abingdon Business Park,Abingdon, Oxon OX14 1DY UNITED KINGDOM
Telephone:[44](1235) 550420 Fax:[44](1235) 550485<German Sales Office>1756 Automation Parkway, San Jose, CA 95131 U.S.A.
Telephone:[1](408) 263-5333 Fax:[1](408) 263-0119

A&D Australasia
ia Ptyy Ltd.
d.
32 Dew Street, Thebarton, South Australia 5031 AUSTRALIA
Telephone:[61](8) 8301-8100 Fax:[61](8) 8352-7409

Hamburger Straße 30 D-22926 Ahrensburg GERMANY
Telephone:[49](0) 4102 459230 Fax:[49](0) 4102 459231

Vereyskaya str.17, Moscow, 121357 RUSSIA
Telephone: [7] (495) 937-33-44 Fax: [7] (495) 937-55-66

A&DD Inst
Instrumentsuments India Pr
Privatee Limited
mited509 Udyog Vihar Phase V
Gurgaon-122 016, Haryana, INDIA
Telephone: [91](124) 471-5555 Fax: [91](124) 471-5599

817, Manhattan Bldg., 33, Gukjegeumyung-ro 6-gil,Yeongdeungpo-gu, Seoul, 07331, KOREA
Telephone:[82](2) 780-4101 Fax:[82](2) 782-4280* SA-ADCC-01-PR1-16301