1WMPD4002746B

This Manual and Marks
All safety messages are identified by the following, “WARNING” or “CAUTION”, of ANSI
Z535.4 (American National Standard Institute: Product Safety Signs and Labels). The
meanings are as follows:

WARNINGA potentially hazardous situation, which if not avoided, could result
in death or serious injury.

CAUTION

A potentially hazardous situation, which if not avoided, may resultin minor or moderate injury.

This is a hazard alert mark.

NoteThis manual is subject to change without notice at any time to improve the product. No
part of this manual may be photocopied, reproduced, or translated into another language
without the prior written consent of the A&D Company.

Product specifications are subject to change without any obligation on the part of the
manufacturer.

Compliance with FCC rules
Please note that this equipment generates, uses and can radiate radio frequency energy.
This equipment has been tested and has been found to comply with the limits of a Class
A computing device pursuant to Subpart J of Part 15 of FCC rules. These rules aredesigned to provide reasonable protection against interference when the equipment is
operated in a commercial environment. If this unit is operated in a residential area it may
cause some interference and under these circumstances the user would be required totake, at his own expense, whatever measures are necessary to eliminate theinterference.
(FCC = Federal Communications Commission in the U.S.A.)

Copyright

2015

Microsoft, Excel and Word are trademarks of Microsoft Corporation.

CONTENTS
1.

INTRODUCTION ....................................................................................................................3

2.

UNPACKING...........................................................................................................................3

3.

PART NAMES AND FUNCTIONS .........................................................................................44.

SETTING UP ..........................................................................................................................6
4.1.
Setting Up Your Balance...............................................................................................6
4.2.
Power Source ...............................................................................................................7
4.3.
Breeze Break ................................................................................................................85.

OPERATION...........................................................................................................................9
5.1.
Turn The Power ON And OFF......................................................................................95.2.
LCD Backlight ...............................................................................................................9
5.3.
Units ............................................................................................................................10
5.4.
Selecting A Weighing Unit...........................................................................................10
5.5.
Basic Operation .......................................................................................................... 11
5.6.
Display Hold Mode...................................................................................................... 115.7.
Counting Mode (pcs) ..................................................................................................12
5.8.
Percent Mode (%).......................................................................................................13

6.

ACCUMULATION FUNCTION.............................................................................................14

7.

CALIBRATION ......................................................................................................................16
7.1.
Calibration Using A Weight .........................................................................................16
7.2.
Gravity Acceleration Correction..................................................................................17

8.

FUNCTIONS.........................................................................................................................18
8.1.
Key Operation.............................................................................................................18
8.2.
Entering The Function Setting Mode..........................................................................18
8.3.
Setting Example..........................................................................................................19
8.4.
Storing Weighing Units ...............................................................................................20
8.5.
Function List................................................................................................................21

9.

OPTIONS..............................................................................................................................23
9.1.
EJ-02 USB Interface...................................................................................................23
9.2.
EJ-03 RS-232C Serial Interface .................................................................................24
9.3.
Data Format ................................................................................................................25
9.4.
EJ-13 Density Determination Kit ................................................................................27

10. ID NUMBER AND GLP.........................................................................................................30
10.1.
Setting The ID Number............................................................................................30
10.2.
GLP Report..............................................................................................................31
11. MAINTENANCE ...................................................................................................................35
11.1.
Notes On Maintenance ...........................................................................................35
11.2.
Error Codes .............................................................................................................35
1

12. SPECIFICATIONS................................................................................................................37
12.1.
Generals ..................................................................................................................37
12.2.
Weighing Units.........................................................................................................37
12.3.
Options ....................................................................................................................38
12.4.
Dimensions..............................................................................................................38
13. GRAVITY ACCELERATION MAP ........................................................................................392

1. INTRODUCTION
This manual describes how this balance works and how to get the most out of it interms of performance.
The EJ-123 and EJ-303 compact precision balance has the following features:
The balance is high-resolution type electronic balance having a display resolution of
1/120,000 to 1/300,000.
The balance has a counting function, a percent function, a hold function and a total
function.
The LCD backlight will help with use in a dimly lighted place.
The balance can be operated with an AC adapter, or 4 x “AA” size dry-cell batteries
for cordless operation.
The optional RS-232C serial interface can be connected with a printer or personal
computer, and Good Laboratory Practice (GLP) data can be obtained.
The optional USB interface is available for connection to a personal computer.

2. UNPACKING
When unpacking the product, check whether all of the following items are included andremove the protective film before use:
Main unit
EJ-123 / EJ-303

AC adapter

Weighing pan

Please confirm that the AC adapter
type is correct for your local voltage
and receptacle typeProtective film

Instruction ManualWind shield cap

33. PART NAMES AND FUNCTIONS
Total indicator

Measurement in air
for specific gravity

M+ indicator
ACAI indicator

Measurement in water
for specific gravity

Battery indicator

Density (Specific gravity)

Hold indicator
STABLE indicator
NET indicator

STABLE

Weighing units

NET
ZERO

ZERO indicator

Wind shield cap(Accessory)

Breeze break

Weighing pan
Spirit level
ON:OFF

key

LCD

Calibration switch cover (bottom)
Leveling foot

SAMPLE

key

PRINT

key

RE-ZERO

key

UNITS

key

EJ-02/EJ-03 slot
Battery cover
Security slot
(Use a security cablecommercially available.)

AC adapter jack4Turns the power ON or OFF.

Switches the weighing units and modes.

Clears the display to zero (combined zero and tare).
Held down to enter the function setting mode.
pcs mode: Enters the sample unit weight storing mode.
% mode:Enters the 100% weight storing mode.
Total mode: Adds the weight value to total.
Outputs the weight value to a printer, or computer.
Enters a unit weight, 100% weight or other setting values to the balance.

54. SETTING UP
4.1.

Setting Up Your Balance

1. Place the weighing pan on the main unit.
2. Adjust the level of the balance using the leveling feet. Use the spirit level to confirm.
The bubble should be in the center of the circle.
3. Calibrate your balance before use. (Refer to “7. CALIBRATION”)

About the wind shield cap
Should the displayed weight value be unstable due to a draft, attach the accessory
wind shield cap to the hole located on the top of the breeze break.
It is possible to reduce the entry of a draft into the breeze break to some extent.
If attached, it may take time for the STABLE indicator to be displayed due to a variancein air pressure as a result of the closure of the breeze break.
Wind shield cap (accessory)

Balance location
To measure correctly, to keep the balance in good condition, and to prevent hazards,
observe the following:
Do not install the balance in locations that are subject to excessive dust, breezes, vibration,
large temperature fluctuations, condensation, or that may have magnetic fields.
Do not install the balance on a surface that is soft or that may cause the balance levelto shift.
Do not install the balance in direct sunshine.
Do not install the balance near heaters or air conditioners.
Do not use an unstable AC power source.
Do not install the balance in a place where combustible or corrosive gases may exist.
Allow the balance to reach equilibrium with the ambient temperature before use.
To ensure accurate weighing, switch the power ON at least half an hour before use, or
have the AC adapter connected to the balance.
When the balance is installed for the first time, or the balance has been moved, carryout calibration as described in “7. CALIBRATION”.

6

4.2.

Power Source

For the power source, the AC adapter or 4 x “AA”
size dry-cell batteries can be used.

When Using The AC Adapter
Use a stable power source. To use the AC adapter,
insert the AC adapter plug into the AC adapter jackAC adapter jackon the rear side of the balance.
The balance is always connected to power internally (warming up status) while the AC
adapter is connected to the balance even when nothing is shown on the display.

When Using The Dry-Cell Batteries
Prepare 4 x AA size (LR6 / R6P) dry cell batteries. The batteries are not included withthe product.

CAUTION
Take great care with the polarity of the batteries. The polarity marks are shown in thebattery box.
Replace used batteries with four new ones when “lb0” is displayed.
Do not mix used and new batteries. Do not mix battery types. It may cause damage tothe batteries or the balance.
The battery life depends on the ambient temperature, how used and so on.
Remove batteries when the balance is not to be used for a long time.
They may leak and cause damage to the balance.
Damage due to battery leakage is not covered by the warranty.

Installing Batteries
1.
2.
3.
4.

Turn off the balance and disconnect the AC adapter if used.
Slide the battery cover off.
Push the battery box up to unhook and pull it out.
Insert new four batteries (LR6 / R6P / AA size) into the battery
box, taking extreme care of the polarities.
5. Push the battery box into the balance as before.
6. Attach the battery cover.
The battery indicator turns on when the balanceis powered by the batteries. It will change as thebattery voltage decreases.
New





Battery coverComing close to “lb0”.

4 x batteries

HookBattery box
7

Hook

4.3.

Breeze BreakThe breeze break can prevent a measurement error due to a (disturbing) wind.
Wipe it with a dry lint free cloth or a lint free cloth which is moistened with warm water.
To keep the antistatic treatment, do not use alcohols and solvents to clean the balance.
When an antistatic treatment becomes weak, apply an antistatic agent to the clear
plastic parts.
The clear cover can be disconnected from the fixing frame easily when adding an
excessive stress during operation. Assemble the breeze break with the following steps.

Assembling the breeze break
1. Slide back the fixing frame toattach the breeze break.
(Or install the fixing frame into theguide on the balance.)

Step 1Fixing frame
Guide
Hook

2. Attach the axis of the breeze
break on the groove of the fixing
frame when keeping the angle of
90 degrees.

Guide
Hook

3. Slide and fix the breeze break.

Breeze break

4. Close the breeze break.
90°
GrooveStep 2

Pivot axis

Breeze break
Step 3

Step 48

5. OPERATION
5.1.

Turn The Power ON And OFF

1. Press the

ON:OFF

key to turn the power ON.

STABLE

Weighing units

NET
ZERO

All of the symbols are displayed as shown above.
(About units: Only the available units will be displayed.)
When the scale is operating using dry-cell batteries, the scale performs count adown using the under-bar on the display.
The display turns off except for a weighing unit and the decimal point.
The balance waits for the weight value to become stable, and then, zero will be
displayed with the ZERO indicator (power-on zero).
The range for power-on zero is within ±50% of the weighing capacity around thecalibrated zero point.
If the power is switched ON while there is a load beyond this range, the balance will betared to zero and the NET indicator and the ZERO indicator turn on.
2. Press the

ON:OFF

key again, and the power will be switched OFF.

Auto-power off function
It is possible to have the power automatically switched OFF, if the zero band
display (band width: ±4 digit from the zero point) continues for approximately 5
minutes. Refer to “8.5. Function List” and set the function to “poff”.

5.2.

LCD Backlight

The LCD backlight will turn on when the weight value changes more than 4d (4 x min.
display division) or any key operation is done. When the weight value becomes andstays stable for some moments, the backlight will automatically turn off. There is also asetting that the backlight is always on or off. For details, see the function setting “ltUp”
of “8.5. Function list”.

9

5.3.

Units

The most common unit of weight used around the world is the gram, but there is often
a need to shift to alternative units specific to the country where the balance is used or
to select modes such as counting or percent.
The units and the order they appear in the display are as follows:
Ounce (avoir)

Troy ounce
Metric carat

Percent mode

Pennyweight

Counting mode
Grain

Gram
NewtonDensity (specific gravity)

Among the units, those available for the user have been set at the factory before
shipping. The unit can be selected in the function setting mode. The order of the units
available is the same as above.
Note
It is possible to store only the units that will be actually used from the units
available. It is also possible to specify the unit that will be shown first when the
power is switched ON. For details, see “8.4. Storing weighing units”.
Conversion table
Units
Name
oz
Ounce (avoir)ozt
Troy ounce
ct
Metric caratdwt
PennyweightGN
Grain (UK)N
NewtonConversion to gram
28.349523125 g
31.1034768 g
0.2 g
1.55517384 g
0.06479891 g(value in Newton) x 1000 / (9.80665 m/s2) g“Newton” is the value calculated by “(weight in grams) x (9.80665 m/s2) / 1000”.

5.4.

Selecting A Weighing Unit

Press theUNITS

key to select a unit.

The following sections are a description of
the three common units: g (gram mode), pcs(counting mode), and % (percent mode).

Each pressing switches the units
available in the order describedon the previous page.
STABLE indicator

ZERO

10

0.000

5.5.

Basic Operation

1. Select a weighing unit.

Container

2. When the display does not show zero, press
the RE-ZERO key to set the display to zero.
Weighing pan

3. When using a tare (container), place the containeron the weighing pan, and press the RE-ZERO
key to subtract the tare weight.
4. Place the object to be weighed on the pan or in
the container.
Wait for the STABLE indicator to be displayed
and read the value.
When using the display hold mode, the holdindicator HOLD is displayed.
5. Remove the object from the pan.
The RE-ZERO key will zero the balance if
the weight is within ±2% of the weighing
capacity around the power-on zero point.
The ZERO indicatorturns on. When theweight exceeds +2% of the weighing
capacity, it will be subtracted from zero as atare weight. In this case the ZERO and NET
indicators turn on.

NET
ZERO

0.000
Object to beweighedSTABLE indicatorNET060.003

Precautions during operation
Make sure that the STABLE indicator is on whenever reading or storing a value.
Do not press the keys with a sharp object such as a pencil.
Do not apply a shock or a load to the pan that is beyond the weighing capacity.
Keep the balance free from foreign objects such as dust or liquid.
Calibrate the balance periodically to maintain weighing accuracy.
(Refer to “7. CALIBRATION”.)

5.6.

Display Hold ModeIn stable measurement, this function can pause the weight value to read it easily. The
function pauses the value except zero band. (band width : ±4 digit from the zero point)
The function releases the value when the weight value changes above the preset value.
Also, when the RE-ZERO key is pressed, the display hold is canceled and display
becomes zero. Refer to the item "Hold" of "8.5. Function List".

11

5.7.

Counting Mode (pcs)

The balance weighs the sample pieces and calculates the unit weight.
Using the sample unit weight, the scale counts the number of items in the sample.
As for the minimum unit weight acceptable, see the function setting “Umin”.
Selecting The Counting Mode
1. Press the UNITS key to select.(0:pieces)

Storing The Sample Unit
2. Press the SAMPLE key to enter the sample unit
weight storing mode.

10 -3. To select the number of samples, press the
1SAMPLE key. It may be set to 5, 10, 25, 50, or 100.
4. Place a tare container on the weighing pan, andpress the RE-ZERO key. Confirm that theright side of the number of samples shows zero.

Switches the number of samples

25 -5. Place the number of samples specified on the pan.
In this example, 25 pieces.

Container

6. Press the PRINT key to calculate and store theunit weight. Remove the sample. The balance is set
to count objects with this unit weight.

Weighing pan

When a unit weight is too light to store, the display
shows lofor the moment and returns to theformer display. When sample weight is light and thecounting error could become large, the balance will
request you to use a larger number of samples.
Place the displayed number of samples on the pan
and press the PRINT key to calculate and store
the unit weight. See also "Sample quantity notice"and "Unit weight error" of "11.2.Error codes".

Confirm the display
ZERO

25 0

Samples(25 pcs)

Counting The Objects
7. Place the objects to be counted on the pan.
Counting Mode Using The ACAI Function
ACAITM (Automatic Counting Accuracy Improvement)
is a function that improves the accuracy of the unit
weight by increasing the number of samples as thecounting process proceeds.
8. After fixing the unit weight at step 6, add a few
more samples on the pan. Then, the ACAI indicator
turn on. (Add more than three pieces. But theindicator will not turn on for the number of samples
beyond the ACAI range.)

To store

25
ACAI indicator
ACAI

25

9. The ACAI indicator will blink and the balance re-calculates the unit weight. Do nottouch the balance or samples on the pan until the ACAI indicator turns off.

12

10. Counting accuracy is improved when the ACAI indicator turns off. Each time the above
operation is performed, a more accurate unit weight will be obtained. There is no
definite upper limit of ACAI range for the number of samples exceeding 100. Try to add
a similar number of samples as that displayed.

5.8.

Percent Mode (%)

Displays the weight value in percentage compared to a reference (100%) weight.
Selecting The Percent Mode
1. Press the UNITS key to select %. (%:percent)

Storing The Reference (100%) Weight
2. Press the SAMPLE key to enter the reference
weight storing mode.

3. Press the

RE-ZERO

0.0

%

100 -

%

100 0

%

key to display 100 0%.
ZERO

4. Place the sample to be set as the reference weighton the pan.

5. Press the PRINT key to store the reference
weight. Remove the sample.
When a unit weight is too light to store, the display
shows
for the moment and returns to theformer display.

Sample corresponding
to the 100% weight

Weighing pan
To store

100.0%

Object to be compared
Reading The Percentage
6. Place the object to be compared to the reference
weight on the pan. The displayed percentage isbased on 100% of the reference weight.

13

75.8

%

6. ACCUMULATION FUNCTION
This function counts the number of times articles are weighed, calculates the total mass
value and can display the number (accumulation count) and accumulated mass value.
Preset the item "Hold" of the function list to use the accumulation function.
The accumulation count and accumulated mass value are stored in the scale even if
the power is removed.

Indicators And Keys
The
indicator is displayed when storing the accumulation data.
The
indicator andis displayed when displaying the accumulated count andaccumulated mass.
Operation keys as follows :The UNITS key
When storing the accumulation data and pressing the UNITS key, the current
weight value, the accumulated count and accumulated mass value are displayed
sequentially.
When the accumulation data is not stored, the UNITS key switches the preset
units sequentially.
The RE-ZERO key
When displaying the weight value and pressing the RE-ZERO key,
zero is displayed.
While displaying the accumulated count and accumulated mass, when pressing
the RE-ZERO key, the accumulation data is reset to the zero.
The SAMPLE key
The condition of the accumulation function is as follows :When a stable weight value is above ±5 digits from zero band.
The accumulated count is less than 1000.
The accumulated mass value is less than 9999999.
The unit weight of the counting mode and the reference (100%) weight of the percent
mode can not store when using the accumulation function.

14

Operations
Selecting A Weighing Unit
1. Select a unit to be used in the accumulation
function with pressing the UNITS key.
The unit can not change after storing theaccumulation data. If changing the unit, resetthe current accumulation data, change the unit.
Plural units can not use at the same time.

ZERO

Weighing pan

ZERO

4. When displaying stable weight value, press the
SAMPLE key to accumulate it. The
indicator is displayed.
5. Repeat step 2, 3, 4 to weighother samples.
The
indicator blinks atonce for each sample.

the

g

Unit

Adding Accumulation Data
2. Press the RE-ZERO key to display zero of
the weighing measurement.
3. Place the sample on the pan.

0.000

0.000

g

60.003g

SampleWhen accumulating data,
indicator is displayed.
M+

Displaying Accumulation Data
6. When pressing the UNITS key,
the current weight value, theaccumulated count andaccumulated mass value aredisplayed sequentially. The
indicator andis displayedwhen displaying the accumulated
count and accumulated mass.

Weight value(
and unit)
Accumulated count(and)

Accumulated mass
( ,and unit)

060.003 g
M+

3

M+

180.009 g

Deleting Accumulation Data
1. Display accumulated count or accumulated mass value.
2. Press the RE-ZERO key to reset the accumulated count and accumulated mass to zero.
The
indicator andare turned off and the state of the balance proceeds tostep 1 on this page.
When the Unit of the function list is changed, the accumulation data are reset.

15

7. CALIBRATIONThis function adjusts the balance for accurate weighing.
Perform calibration in the following cases.
When the balance is first used.
When the balance has been moved.
For regular calibration.
Calibration (CAL) switch
Calibration (CAL) switch cover

7.1.

Calibration Using A Weight

Prepare a calibration weight (sold separately) in advance.
1. Warm up the balance for at least half an hour withnothing on the pan.
2. Press and hold the calibration (CAL) switch until
Cal
appears, and release the switch.
3. The balance displays Cal 0 .
To change the calibration weight value, proceed to step 4.
To use the calibration weight value in the balance
memory, proceed to step 5.
4. Press the SAMPLE key. The display shows thecalibration weight value in “grams” that is stored in thebalance. Use the following keys to change the value.
SAMPLE key ......To select the digit blinking to be changed.

Press and holdthe CAL switch

Cal
Release the CAL switch

Cal 00100.000
Set the weight using
the relevant keys

RE-ZERO key ....To set the value of the digit selected.
PRINT key...........To store the value and return to step 3.
UNITS key...........To cancel the value and return to step 3.
Use a calibration weight more than 2/3 of the capacity.

Cal 0
With nothing
on the pan

5. At step 3, press the PRINT key to weigh thezero-point value. Do not touch the pan during weighing.
When the zero calibration is completed, the display willshow the calibration weight value.
To perform the zero calibration only and finish theprocedure, press the UNITS key. After displaying
end, the balance turns off.
6. Place a calibration weight with the same value as displayedon the pan.
Press the PRINT key to weigh it. Do not touch the panduring weighing.
16

To store

Cal 0
100.000
Calibration
weight7.

end
is displayed.
Remove the weight from the pan.

To store

When the GLP output (function setting “info 1” or“info 2”) is selected, glp
is displayed. The
calibration report is output and end
isdisplayed again. Then the balance will be turned off.
Note
The value set in step 4 is stored in memory evenafter the power is switched OFF.
If the balance is to be moved to other places, set
the gravity acceleration value of the area where
the calibration using a weight is to be done, andcalibrate the balance according to the procedure
above. See the next section to set the value.

7.2.

100.000
0end000
GLP output

glp

end000The balance is turned off.

Gravity Acceleration Correction

When the balance is first used or has been moved to a different place, it should becalibrated using a calibration weight. But if a calibration weight is not available, thegravity acceleration correction will compensate the balance. Change the gravity
acceleration value of the balance to the value of the area where the balance will be
used. See the gravity acceleration map appended to the end of this manual.
Note
Gravity acceleration correction is not required when
the balance is calibrated using a calibration weightat the place where the balance is to be used.

1. Press and hold the calibration (CAL) switch until
Cal
appears, and release the switch.
2. The balance displays

Cal 0

.

3. Press the RE-ZERO key. The display shows thegravity acceleration value stored in the balance. Use thefollowing keys to change the value.

Press and holdthe CAL switch

Cal
Release the CAL switch

Cal 0

SAMPLE key ......To select the digit blinking to be changed.
RE-ZERO key ....To set the value of the digit selected.
PRINT key...........To store the value and return to step 2.
UNITS key...........To cancel the value and return to step 2.

9.7985
Set the value using therelevant keys

4. After setting the value, press the PRINT key.
end
and Cal 0
is displayed again.
5. If it is necessary to calibrate the balance using a calibration
weight, go to step 4 of "7.1. Calibration Using A Weight".
To finish the setting, press the UNITS key.
6. The balance is turned off.

To storeend
Cal 0
The balance is turned off.

17

8. FUNCTIONS
8.1.

Key Operation
The ON:OFF key to cancel the operation and turns
the balance off.
Item key to select a class and an item.
In the weighing mode, press and hold the key to enter
the function setting mode.
The enter key to proceed to the selected class andstore the setting per class and goes to the next class.
The parameter key to select a parameter.
The cancel key to cancel the operation, and proceed to
the next class or return to the weighing mode.

8.2.

Entering The Function Setting ModeIn the weighing mode, press and hold the SAMPLE key
to enter the function setting mode and display func .
Each time the SAMPLE
key is pressed, the class appears one after another.
Once a class is selected, the items areavailable for selection. (Refer to “8.5. Function List”.)

Press and holdfunc

Each pressing switches the class

Unit
•
•
•
Set using therelevant keys

18

8.3.

Setting Example

To set auto power-off function to “Enabled”, and the ACAI function to “Disabled”.
1. Press and hold the

SAMPLE

key to display

func

.

Press and holdfunc
2. Press the

PRINT

key. The balance displays

poff 0 .

poff 03. Press the

RE-ZERO

key to displaypoff 1 .
Each pressing switches
the parameter

4. Press the SAMPLE
aCai 1 .

key several times to displaypoff 1
To confirm

aCai 1
5. Press the

RE-ZERO

key to selectaCai 0 .
Each pressing switches
the parameter

6. Press the PRINT key to store the parameters.
Unitappears after end.

7. Press the

UNITS

key to turn off the balance.

aCai 0To store

end
Unit
The balance returns off.

19

8.4.

Storing Weighing Units

It is possible to store the weighing units that will actually be used from the units
available. For the units available, see “5.3. Units”.
Select and store the weighing units as described below:
1. Press and hold the

SAMPLE

2. Press the

SAMPLE

3. Press the

PRINT

func

key to display

key to display

Unit

.

func
Unit

key.

4. Press the SAMPLE
be stored.

Press and hold.

key to display a weighing unit to

Unit

Each pressing switches the units available
in the order described on "5.3.Units".
5. Press the RE-ZERO key to select the weighing unit. The
selected weighing unit is shown with the STABLE indicator.

Unit

At this stage, the weighing unit is not stored in memory.

To select
To store

6. Repeat steps 4. and 5. to store all weighing units to be used.

7. Press the PRINT key to store the selected weighing
units in memory. id
is displayed after end
.

end
id

8. Press the

UNITS

key to turn off the balance.

Note
When the balance is switched ON, it starts with the unit thatwas stored first at step 5.

20

The balance is turned off.

8.5.

Function List

ParamDescriptioneter 0 Auto power-off disabledpoff
Automatically power off
Auto power-off
1 Auto power-off enabled
0 Fast / sensitive
Cond
Software filtering 1
Response
2 Slow / stable
The STABLE indicator is
0 Judge precisely (± 0.5d)
displayed when the fluctuation
5t-b(± 1d)
 1
interval of a regular time
Stability band width
weighing display is less than
2 Judge moderately (± 2d)
the reference value.
Function to keep the
0 Disabledtrc
display value at zero by
Zero tracking
 1 Enabledtracking the zero point
 0 Point (.)
pnt
Decimal separator
Decimal point
1 Comma (,)
0 Command and stream modes 1 Command and PRINT key
prt
PRINT key andAuto-print A: + data
2 Command,
Data output
auto-print A
Auto-print B: +/- data
mode
PRINT key and
3 Command,
auto-print B
4 Command mode only
pause (general 0 NopU5eInterval between
equipment)
func
continuous data
Data output pause
1 1.6 seconds (for AD-8121)
 0 No output
info
GLP output format
1 AD-8121 format (*)
GLP output
2 General format
 0 2400 bps
1 4800 bps
bp5
Baud rate
2 9600 bps
3 1200 bps
 0 7 bits, even parity
btpr
1 7 bits, odd parity
Data and parity
2 8 bits, non parity
0 ACAI disabled
If “0” is set, no additionalaCaisamples required.
ACAI function
 1 ACAI enabled
 0 1dUmin
d = digit : in unit of
1 1/10 d
minimum display
Minimum unit weight
2 total sample weight 5d (**)
 0 10 pcs
1 25 pcs
5mpl
The number of samples shown first when
2 50 pcs
entered into the unit weight storing mode
Sample number
3 100 pcs
4 5 pcs
 : Factory setting

Class

Item

21

ParamDescriptioneter 0 Water temperatureldin
The way to input liquid density.
Liquid density input
1 Liquid density
 0 Not usedd = digit : in unit of minimum display
1 Hold. Limit : ±10 d. Hold: When a stable value leaves
Hold
2 Hold. Limit : ±50 d. more than 4 digits from zero band,
Display hold function
the display value is held. When the
3 Hold. Limit : ±100 d. value is change more than a limit,
4 Hold. Limit : ±200 d. the display hold is released.
func totl
 0 Not used
Accumulation function
1 Accumulated
0 Always off
1 Turns off after 5 seconds
To control how the LCD
2
Turns
off
after
10
seconds
ltUpbacklight turns off. Weight
LCD Backlight control  3 Turns off after 30 seconds change or key operation will
4 Turns off after 60 seconds turn the backlight on.
5 Always on
Weighing units to be
Refer to “8.4. Storing
Unit
Sets to display units
displayed
Weighing Units”
Refer to “10. ID NUMBER
id ID number for GLP output Sets the ID number
AND GLP”
 : Factory setting

Class Item

* : When the AD-8121 format is selected, the interval between data is 1.6 seconds
regardless of the setting “pU5e”.
** : Even if the weight display is “5d”, the sample weight may not be accepted. This isbecause the weight display data is rounded off internally.

22

9. OPTIONS
The following options are available for the EJ-123/EJ-303 balance:
EJ-02
USB interface
EJ-03
RS-232C serial interface
EJ-13
Density determination kit
Note
The EJ-123/EJ-303 balance have only one option slot for a communications
interface. Either the EJ-02 USB interface or the EJ-03 RS-232C serial interface can beinstalled at one time.

9.1.

EJ-02 USB Interface

The EJ-02 allows an EJ-123/EJ-303 balance to be connected with a printer or apersonal computer with the USB interface.
The EJ-02 can transmit the weight value (numerical value only) uni-directionally to apersonal computer via USB.
The EJ-02 can transmit the weight value (numerical value only) directly to application
software such as Microsoft Excel, Word, memo pad, and so on.
A special USB software driver is not necessary.
The EJ-02 cannot be used for bidirectional communication.

EJ-02 Installation
1. Turn off the balance and disconnect the AC
adapter if used.

Option slot for
EJ-02 / EJ-03

2. Remove the cover of the option slot on the rearby pressing and lowering it down.

Push down andremove the cover

3. Connect the connector in the slot to the EJ-02
unit and insert it into the slot.

EJ-02

Step 2

4. Secure the EJ-02 with the screws supplied withthe option.
Step 3Connector inside
the option slot

23

EJ-02

9.2.

EJ-03 RS-232C Serial Interface

The EJ-03 allows an EJ-123/EJ-303 balance to be connected with a printer or apersonal computer with the RS-232C interface.
The RS-232C serial interface has the following four modes.
Stream mode ...................Outputs data continuously.
Key mode..........................Outputs data by pressing the PRINT key.
Auto-print mode .............Outputs data which meets the conditions of auto-print.
Command mode ............Controls the balance using commands from a computer.
Set the parameters of the data format (bp5 and btpr) and data output mode (prt),
as necessary.
Use a D-Sub 9 pin cable (straight type) to connect with a computer.
Optional cable: AX-KO2466-200 D-Sub 9 pin / 9 pin cable (2 m long)

EJ-03 Installation
1. Turn off the balance and disconnect the AC
adapter if used.
2. Remove the cover of the option slot on the rearby pressing and lowering it down.
3. Connect the connector in the slot to the EJ-03
unit and insert it into the slot.
4. Secure the EJ-03 with the screws supplied withthe option.

Option slot for
EJ-02 / EJ-03
Push down andremove the cover

Step 2

EJ-03

Step 3

EJ-03

Connector inside
the option slot

Interface Specifications
Transmission system
Transmission form
Data format

LSB
0

1

EIA RS-232C
Asynchronous, bi-directional, half-duplex
Baud rate: 1200, 2400, 4800, 9600 bps
Data:
7 bits + parity 1bit (even or odd) or 8 bits (non-parity)
Start bit:
1 bit
Stop bit:
1 bit
Code:
ASCII
Terminator: CRLF (CR: 0Dh, LF: 0Ah)
2

3

4

5

MSB
6

1 (-15V ~ -5V)
0 (+5V ~ +15V)
Stop bit
Parity bit
Data bit
Start bit24

Pin Connections
D-Sub 9 pin male connector
1
2
3
4
5
6
7
8
9

Inside of the balance

N.C.
Transmit data
Receive data
N.C.
Signal ground
Data set ready
Request to send
Clear to send
N.C.

The interface is designated as DCE (Data Communication Equipment).

9.3.

Data Format
S

T

,

+

Header

0

0

0

0

.

0

0

0

g CR LF

Data
Separator

Unit
Terminator(“ ” shows a space.)

There are four types of headers:ST :Stable weighing data (including % data)
QT :Stable counting data
US :Unstable weighing data (including counting and % data)OL :Out of weighing range (Over)The data is normally 9 digits including a decimal point and a sign.
There are 10 types of units:g : Weighing data “gram”
P C : Counting data “pcs”% : Percentage data “%”o z : Weighing data “decimal ounce”
o z t : Weighing data “troy ounce”
c t: Weighing data “carat”
d w t : Weighing data “penny weight”
GN : Weighing data “grain”
N : Force data “Newton”
D S : Calculated density (specific gravity) value
The terminator is always CRLF.
Example of output data:
Weighing data “gram”

S T , + 0 0 1 2

3 4 5

g CR LF

Counting data

Q T , + 0 0 0 1 2 3 4 5

P C CR LF

Percentage data

S T , + 0 0 0 1 2 3Out of range “gram” (+)

O L, + 9 9 9 9Out of range “pcs” (-)

O L

,
25

.

4

% CR LF

9 9 9

g CR LF- 9 9 9 9 9 9 9 9

P C CR LF.

.

Data Output Mode
Stream mode
Set the function to “prt 0”.
The balance outputs the current display data. The data-update rate is approximately 10
times per second. This rate is the same as the display-update.
The balance does not output data while it is in the setting mode.
Key mode
Set the function to “prt 1, 2 or 3”.
When the PRINT key is pressed while the weight value is stable (the STABLE
indicator is on), the balance transmits the data. When the data is transmitted, thedisplay will blink one time.
Auto-print mode A
Set the function to “prt 2”.
The balance transmits the weight value when the display is stable (the STABLE
indicator is on) and the data is greater than +4 digits.
The next output can be obtained after the display returns below +4 digits.
Auto-print mode B
Set the function to “prt 3”.
The balance transmits the weight value when the display is stable (the STABLE indicatoris on) and the data is greater than +4 digits or less than -4 digits.
The next output can be obtained after the display returns between -4 digits and +4 digits.

Command ModeIn the command mode, the balance is controlled by commands that come from thepersonal computer and so on.
Command List
Command to request the current weighing data.
Command

Q CR LF

Reply

S T , + 0 0 1 2

.

3 4 5

Command to zero or tare the balance (same as theCommand

Z CR LF

Reply

Z CR LF

RE-ZERO

Command to change the weighing units (same as theCommand

U CR LF

Reply

U CR LF

26

g CR LF

UNITS

key).

key).

9.4.

EJ-13 Density Determination Kit

Using this option and calculation program,the balance can determine the density(specific gravity) of a sample.

Breeze break
Weighing pan

EJ-13 Installation
Remove the weighing pan and the fixing
frame.
Install the beaker stand on the balance
and attach the pan stand to the pan
support of the balance.

Fixing frame

Remove the breeze break that can not be
used while using the density
determination kit.

Sample weighing pan
Upper pan to weigh in air
Pan standPlace a beaker filled with water on the
beaker stand and place the sample
weighing pan on top of the pan stand.

Lower pan to weigh in water
Beaker standPan support

Density (Specific Gravity) Measurement
The density for a liquid can be set two ways.
One is to set the water temperature and theother is to set the density value directly.
The factory setting for density of a liquid is
25 °C as water temperature (the density
value, ρ = 0.99704 (g/cm3), is used for
calculation).

A : To weigh A in air.

The result is shown with three decimal places.
The density (specific gravity) is calculated by
the following formula.
AS=xρ
S: Density (specific gravity) of a sample
A-B
A: Weight in air
B: Weight in liquid
ρ: Density of liquid (water)

L : To weigh B in water.

Change the function table
Selecting a way to set the density of a liquid
Select the liquid density input method from the function table below.
The function table is available only when the density measurement mode is selected.
Parameter 0 Water temperature
ldin
func
Liquid density input
1 Liquid density
 : Factory setting

Class

Item

27

Description
The way to input liquid density.

Entering the density of a liquid (ldin = 0)
1. Press the

UNITS

key to selectSG indicatorSG .
A

2. Press and hold the UNITS key to display the liquiddensity input mode. The display will show the watertemperature currently set (factory setting: 25 °C).

SG

0.000

3. Using the RE-ZERO (to increment the value) andSAMPLE keys (to shift the selected digit), set thevalue and press the PRINT key to store.

SGt---25

To cancel the setting procedure and return to thedensity measuring mode, press the UNITS key.
The input value is not stored.
The relation between the water temperature anddensity is shown below.
°C
0
10
20
30
40
50
60
70
80
90

+0
0.99984
0.99970
0.99820
0.99565
0.992220.98804
0.983200.97777
0.971800.96532+10.999900.99961
0.99799
0.99534
0.99183
0.98758
0.98268
0.977200.971170.96465+20.99994
0.99949
0.99777
0.99503
0.991440.987120.98216
0.976620.970540.96397+3
0.999960.99938
0.997540.99470
0.99104
0.98665
0.98163
0.97603
0.96991
0.96328+4
0.99997
0.99924
0.99730
0.99437
0.99063
0.98618
0.981100.975440.96927
0.96259+50.999960.999100.99704
0.994030.990210.985700.98055
0.974850.96862
0.96190+60.99994
0.99894
0.99678
0.99368
0.98979
0.985210.98001
0.974250.96797
0.96120+70.999900.99877
0.99651
0.99333
0.989360.98471
0.97946
0.973640.967310.96050+80.99985
0.99860
0.99623
0.99297
0.988930.984220.978900.97303
0.966650.95979+90.99978
0.998410.99594
0.99259
0.98849
0.98371
0.97834
0.972420.96600
0.95906

Entering the density of a liquid directly (ldin = 1)
1. Press the

UNITS

key to selectSG .

2. Press and hold the UNITS key to display the liquiddensity input mode. The display will show the liquiddensity currently set (factory setting: 1.0000 g/cm3)
3. Using the RE-ZERO (to increment the value) andSAMPLE keys (to shift the selected digit), set thevalue and press the PRINT key to store.
To cancel the setting procedure and return to thedensity measuring mode, press the UNITS key. The
input value is not stored.

28

SG

d 1.0000

Example of density measurement
Selecting the density measurement mode
1. Press the

UNITS

key to selectSG

Measuring weight in air.

.

SG indicator

The weighing unit is “g”.
The display shows that A blinks and thebalance is measuring the weight in air.

A

SG

A

SG

0.000

When the display does not show zero, press
the RE-ZERO key to set the display to zero.
2. Place a sample on the upper pan.

20.0003. Wait for the STABLE indicator to be displayed andpress the SAMPLE key to store the weight in air.
4. The display shows that L blinks and the balancestarts to measure the weight in water.

Measuring weight in water.
L

SG

L

SG

20.0005. Place the sample on the lower pan in water.
Adjust the amount of water so that the sample isabout 10 mm below the water surface.

17.510

6. Wait for the STABLE indicator to be displayed andpress the SAMPLE key. Then the balance reads
the weight in water and shows the density (specificgravity) of the sample.

SG

7. To continue the density measurement, press
the SAMPLE key again. To exit thismeasurement, press the UNITS key.

A

: To weigh the sample in air.

L: To weigh the sample in water.

29

8.008

10. ID NUMBER AND GLPThe ID number is used to identify the balance when Good Laboratory Practice (GLP) is
used.
The ID number is stored in memory even if power to the balance is switched OFF.
The following GLP data is transmitted to a printer or a computer using the optional
RS-232C serial interface.
The result of calibration (“Calibration report”)
The result of calibration test (“Calibration test report”)
The “Title block” and “End block” for GLP data
The GLP output format includes the balance manufacturer name, model number, serial
number, ID number and space for a signature.
By using the AD-8121B, printing out dates and times is possible (GLP output format
info = 1).

10.1. Setting The ID Number
1. Press and hold the SAMPLE key to display func
2. Press the SAMPLE key several times to display id
3. Press the PRINT key. Enter the ID number using thefollowing keys.
SAMPLE key .......... To select the digit blinking to bechanged.
RE-ZERO key ........ To set the character of the digit
selected. See the table below for the“display character set”.
PRINT key ............... To store the value and proceed to
the next step.
UNITS key ............... To cancel the value and proceed to
the next step.
4. When the above operation has been completed, func
appears after end.
5. Press the UNITS key to turn off the balance.

Press and hold

.
.

func
Press several
times

id
000000Set using the relevant keys

end idfunc
The balance is turned off.

Display Character Set
0 1 2 3 4 5 6 7 8 9 A B C D E F GH I J K L MNO P QR S T U VWX Y Z
0 1 2 3 4 5 6 7 8 9 - _ a b C d e f g H i j k l m n o p q r s t U v w x y z“ ” : Space
30

10.2. GLP Report
To print the GLP report, set the function to “info 1”, “pU5e 1” and set theAD-8121B printer to MODE 3.
To output the GLP report to a personal computer, set the function to “info 2”,“pU5e 0” .

Calibration Report
Perform calibration using a weight. Then the balance will output a calibration report.
Refer to “7.1. Calibration Using A Weight” about the calibration.
1. Perform calibration according to “7.1. Calibration
Using A Weight”.
2.

end
appears when the calibration has
been completed.

3.

glp
output.

4.

Operation of Calibration(See “7.CALIBRATION”)

end

is displayed and calibration report isend
is displayed again. Remove theweight. The balance is turned off.

GLP output

glp
end

The balance is turned off.
AD-8121 format “info 1”___________A_&_D
MODEL_____EJ-123
S/N____Q12345678
ID________ABCDEF
DATE__2013/11/14
TIME____02:53:21
CALIBRATED(EXT.)
CAL.WEIGHT_____+100.000__g
SIGNATURE

General format “info 2”
Manufacturer
Model
Serial number
ID number
Date
Time
Calibration type
Calibration weight
Signature-_-_-_-_-_-_-_-_~ : Space, ASCII 20h
CR : Carriage return, ASCII 0Dh
LF : Line feed, ASCII 04Ah

31~~~~~~~~~~~ A ~ & ~ D <CRLF>
MODEL~~~~~EJ-123<CRLF>
S/N~~~~Q12345678<CRLF>
ID~~~~~~~~ABCDEF<CRLF>
DATE<CRLF><CRLF>
TIME<CRLF><CRLF>CALIBRATED(EXT.)<CRLF>
CAL.WEIGHT<CRLF>~~~~~+100.000~~g<CRLF>
SIGNATURE<CRLF><CRLF><CRLF>----------------<CRLF><CRLF><CRLF>

Calibration Test Report
The calibration test mode is used to compare a calibration test weight with thecalibration test data weighed by the balance.
This test does not perform calibration and this mode is available even after sealing thecalibration (CAL) switch.
Press and hold both keys.
1. Press and hold the SAMPLE and PRINT keys.
CCwill appear. Then, release both keys.
 Pressing and holding the calibration (CAL) switch will
also display CC
after Cal. Then, release
the CAL switch while showing CC
.
2.

CC 0

CC

Release both keys.

CC 0

is displayed.

3. If necessary, change the value of calibration test
weight according to the procedure step 4 of “7.1.
Calibration Using A Weight”.
4. With nothing on the pan, press the PRINT key. The
zero point is measured and the weight value isdisplayed for a few seconds. Then, the display shows
the value of calibration test weight.
5. Place a weight of the same value as displayed on the
pan and press the PRINT key to weigh it. The
weight value is displayed for a few seconds.

0.000
100.000
To weigh

100.001

6.

end

7.

glp
is displayed and calibration test
report is output.

end

8.

end
appears again.
Remove the weight.
The balance is turned off.

glp

appears.

To output calibration test reportend
The balance is turned off.

32

AD-8121 format “info 1”___________A_&_D
MODEL_____EJ-123
S/N____Q12345678
ID________ABCDEF
DATE__2013/11/14
TIME____02:53:21
CAL.TEST(EXT.)ACTUAL________0.000__g_____+100.001__g
TARGET_____+100.000__g
SIGNATURE

General format “info 2”
Manufacturer
Model
Serial number
ID number
Date
Time
Calibration test type
Zero value
Target weight value
Target weight
Signature-_-_-_-_-_-_-_-_~ : Space, ASCII 20h
CR : Carriage return, ASCII 0Dh
LF : Line feed, ASCII 04Ah~~~~~~~~~~~ A ~ & ~ D <CRLF>
MODEL~~~~~EJ-123<CRLF>
S/N~~~~Q12345678<CRLF>
ID~~~~~~~~ABCDEF<CRLF>
DATE<CRLF><CRLF>
TIME<CRLF><CRLF>CAL.TEST(EXT.)<CRLF>
ACTUAL<CRLF>~~~~~~~~0.000~~g<CRLF>~~~~~+100.001~~g<CRLF>
TARGET<CRLF>~~~~~+100.000~~g<CRLF>
SIGNATURE<CRLF><CRLF><CRLF>----------------<CRLF><CRLF><CRLF>

Output Of “Title Block” And “End Block”
When a weight value is recorded as the GLP report. “Title block” and “End block” areadded at the beginning and at the end of a group of weight values.

0.000“Title Block”
1. Press and hold the PRINT key.
Release the PRINT key when
5tart
is displayed. Thebalance outputs the “Title block”.

Press and hold

2. The balance can output theweight value by pressing the
PRINT key or selecting theauto-print mode.

To output “Title block”

5tart
end
12.345•

To output weight value •••23.45633

“End Block”
3. Press and hold the PRINT key.
Release the PRINT key when
recendis displayed. Thebalance outputs the “End block”.
4.

Press and holdrecendTo output “End block”

end
is displayed.
The balance return to the weighing
mode.

end
The balance return to the weighing mode.

AD-8121 format “info 1”___________A_&_D
MODEL_____EJ-123
S/N____Q12345678
ID_______ABCDEF
DATE__2013/11/14
START
TIME____02:53:21

Title block
Manufacturer
Model
Serial number
ID number
Date
Start time

ST,+0012.345__g
ST,+0023.456__g
ST,+0034.567__g
ST,+0045.678__g
END
TIME____02:55:21
SIGNATURE-_-_-_-_-_-_-_-_

Weight value

End time
Signature

End block~ : Space, ASCII 20h
CR : Carriage return, ASCII 0Dh
LF : Line feed, ASCII 04Ah

34

General format “info 2”~~~~~~~~~~~ A ~ & ~ D <CRLF>
MODEL~~~~~EJ-123<CRLF>
S/N~~~~Q12345678<CRLF>
ID~~~~~~~~ABCDEF<CRLF>
DATE<CRLF><CRLF>START<CRLF>
TIME<CRLF><CRLF><CRLF>
ST,+0012.345~~g<CRLF>
ST,+0023.456~~g<CRLF>
ST,+0034.567~~g<CRLF>
ST,+0045.678~~g<CRLF><CRLF>
END<CRLF>
TIME<CRLF><CRLF>
SIGNATURE<CRLF><CRLF><CRLF>----------------<CRLF><CRLF><CRLF>

11. MAINTENANCE
11.1. Notes On MaintenanceDo not disassemble the balance. Contact your local A&D dealer if your balance needs
service or repair.
Please use the original package for transportation.
Do not use organic solvents to clean the balance. Use a lint free cloth dampened withwarm water and a mild detergent.
To clean around the pan support, refer to the illustrations below.
Sample weighing pan
Breeze break
Weighing pan

Upper pan to weigh in air
Pan standLower pan to weigh in water
Beaker standPan support
Hook

Pan support

Fixing frame
Hook

Leveling foot11.2. Error Codes
Overload error

eeeeeeWarning to indicate that an object beyond the balance
capacity has been placed on the pan. Remove the object
from the pan.

Range over notice--eThis will be shown if the weight sensor receives strong force
upward. Check if there is anything sandwiched around theweighing pan. There is a possibility that the weight sensor
itself may have a failure.

Unit / 100% weight errorlo

The sample weight is too light to set the unit weight in the
counting mode or 100% reference weight in the % mode.

Sample quantity notice

10 25 -

When sample weight is light and the counting error could
become large, the balance will request you to use larger
number of samples. Place the displayed number of samples
on the pan and press the PRINT key to store the unit
weight.
35

50 100 -Note: Pressing thePRINT
key without adding
samples may reduce the counting accuracy.
Starting from the 100 samples, 100 - may bedisplayed when the sample weight is light. In thiscase, press the PRINT key without adding any
samples.
When “aCai 0” (ACAI disabled) or “Umin 2” isselected, this notice is not shown.

CAL errors-Cal eWarning to indicate that calibration has been canceled
because the calibration weight is too light.
Check the weighing pan and the calibration weight.

Battery error

lb0
Hb0

Warning to show that the batteries are exhausted.
them with new one or use the AC adapter.

Replace

Warning to show that the battery voltage is too high. Checkthe battery.

AC adapter error

Hb1
lb1

Warning to show that the output voltage of an AC adapter is
too high. Check if the AC adapter is the correct type.
Warning to show that the output voltage of an AC adapter is
too low. Check if the AC adapter is the correct type.

Stability error

error1

Warning to indicate that the weight value is not stable and thebalance cannot display it. Prevent vibration and drafts. Press
the UNITS key to return to the weighing mode.

Internal errors (# is number except 1.)
The balance detects error state in the internal processing.
With nothing on the weighing pan, turn the power off and on
again. If the error persists, request service.

error#

If you cannot cancel an error or other errors have occurred, request service fromthe store where you purchased the balance or from your local A&D dealer.

36

12. SPECIFICATIONS
12.1. Generals
MODEL
Weight capacity
Minimum display “d” = digit
Repeatability (Standard deviation)
Linearity
Sensitivity drift
No. of samples
Max. count *Min. unit weight *Min. % display
Min. 100 % weight
Display
Display update
Operating temp.
Power supply
Battery operation
Weighing pan size
WeightCalibration weight (factory setting)

EJ-123
EJ-303120 g
310 g
0.001 g0.001 g0.003 g0.003 g±0.003 g±0.003 g±20 ppm / °C (10 °C to 30 °C / 50 °F to 86 °F)
5, 10, 25, 50 or 100 pieces
120,000 pcs
310,000 pcs
0.001 g0.01 %
0.1 g
7 segment LCD display with backlight(Character height 16 mm)
10 time per second
10 °C to 30 °C / 50 °F to 86 °F,
less than 85 %R.H. (non-condensing)AC adapter or 4 x “AA” size dry-cell batteries
Approximately 80 hours(backlight off, alkaline batteries)
110 mm ø
Approximately 1160 g
100 g
300 g

* : In case of “Umin 0” (factory setting)

12.2. Weighing Units
oz
ozt
ct
dwt
GN

MODEL
Capacity
Min. display
Capacity
Min. display
Capacity
Min. display
Capacity
Min. display
Capacity
Min. display

EJ-1234.2329 oz
0.0001 oz
3.8581 ozt
0.0001 ozt
600.000ct
0.005
ct
77.162 dwt
0.001 dwt
1851.88
GN
0.02
GN

37

EJ-30310.9349 oz
0.0001 oz
9.9667 ozt
0.0001 ozt
1550.000
ct
0.005
ct
199.335 dwt
0.001 dwt
4784.04
GN
0.02
GN12.3. Options
EJ-02
EJ-03
EJ-13

USB interface
RS-232C serial interface
Density determination kit

12.4. Dimensions
190

Unit: mm

211
133

90

110

○

5
126
Minimum38

13. GRAVITY ACCELERATION MAP
Values Of Gravity At Various Locations
Amsterdam
AthensAuckland NZ
BangkokBirmingham
Brussels
Buenos Aires
Calcutta
Cape Town
ChicagoCopenhagen
CyprusDjakarta
Frankfurt
Glasgow
HavanaHelsinki
KuwaitLisbonLondon (Greenwich)
Los Angeles
Madrid

9.813 m/s2
9.807 m/s2
9.799 m/s2
9.783 m/s2
9.813 m/s2
9.811 m/s2
9.797 m/s2
9.788 m/s2
9.796 m/s2
9.803 m/s2
9.815 m/s2
9.797 m/s2
9.781 m/s2
9.810 m/s2
9.816 m/s2
9.788 m/s2
9.819 m/s2
9.793 m/s2
9.801 m/s2
9.812 m/s2
9.796 m/s2
9.800 m/s2

ManilaMelbourne
Mexico City
Milan
New York
Oslo
OttawaParis
Rio de Janeiro
Rome
San Francisco
Singapore
Stockholm
SydneyTaichung
TainanTaipeiTokyo
Vancouver, BC
Washington DC
Wellington NZ
Zurich39

9.784 m/s2
9.800 m/s2
9.779 m/s2
9.806 m/s2
9.802 m/s2
9.819 m/s2
9.806 m/s2
9.809 m/s2
9.788 m/s2
9.803 m/s2
9.800 m/s2
9.781 m/s2
9.818 m/s2
9.797 m/s2
9.789 m/s2
9.788 m/s2
9.790 m/s2
9.798 m/s2
9.809 m/s2
9.801 m/s2
9.803 m/s2
9.807 m/s2

World Map40

3-23-14 Higashi-Ikebukuro, Toshima-ku, Tokyo 170-0013, JAPAN
Telephone: [81] (3) 5391-6132
Fax: [81] (3) 5391-6148

A&D ENGINEERING, INC.
1756 Automation Parkway, San Jose, California 95131, U.S.A.
Telephone: [1] (408) 263-5333
Fax: [1] (408)263-0119

A&D INSTRUMENTS LIMITED
Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire
Telephone: [44] (1235) 550420
Fax: [44] (1235) 550485

OX14 1DY

United Kingdom

A&D AUSTRALASIA PTY LTD
32 Dew Street, Thebarton, South Australia 5031, AUSTRALIA
Telephone: [61] (8) 8301-8100
Fax: [61] (8) 8352-7409

A&D KOREA Limited

한국에이.엔.디(주)서울특별시 영등포구 국제금융로6길33 (여의도동) 맨하탄빌딩 817 우편 번호 150-749( 817, Manhattan Bldg., 33. Gukjegeumyung-ro 6-gil, Yeongdeungpo-gu, Seoul, 150-749 Korea )전화: [82] (2) 780-4101
팩스: [82] (2) 782-4280

OOO A&D RUS

OOO "ЭЙ энд ДИ РУС"121357, Российская Федерация, г.Москва, ул. Верейская, дом 17( Business-Center "Vereyskaya Plaza-2" 121357, Russian Federation, Moscow, Vereyskaya Street 17 )тел.: [7] (495) 937-33-44
факс: [7] (495) 937-55-66

A&D INSTRUMENTS INDIA PRIVATE LIMITED( 509, Udyog Vihar, Phase- , Gurgaon - 122 016, Haryana, India ): 91-124-4715555: 91-124-4715599