Multi-functional Precision Balances

GX/GFSeries

GX-400(with optional glass breeze break)

GX-600(with standard breeze break)ISO
9000 CERTIFIED

http://www.aandd.jp

The World’s Most User-Friendly
Balances…
* Patent Approved

Internal Calibration
The GX internal calibration with an internal mass saves time and money. External calibration
requires a great deal of skill and time as well as an expensive external mass. With our internal
calibration with internal mass, the balance can be calibrated very quickly.
■ Automatic self calibration – Self-calibrates automatically when the balance detects ambient

temperature changes■ One touch automatic calibration – Allows you to calibrate the balance on demand with justthe push of a button

Quick USB Interface (GX-02)
Easy-to-use optional USB interface, with no driver or software installation necessary, for transfer ofweighing data to a computer (Unidirectional communication from a balance to a PC only. Please use
the standard RS-232C interface for sending commands from a PC to the balance.)123

9Memory

6Internal
Calibration

External
Calibration

(GX )

(GF )

Super HybridSensor

VFD Display

RS-232CInterface

GLP
Compliant

Clock & Calendar(GX )Data Memory
Function

Countingwith ACAI

Percentage
Function

VFD – Vacuum Fluorescent Display • • • • • • • • • •Large VFD is clearly visible even in a poorly-lit workplace.

GLP/GMP/GCP/ISO Compliant • • • • • • • • • • • • • • • • • • • • • • • •In accordance with GLP and LIMS regulations, the balance ID number andcalibration data can be printed out using A&D’s AD-8121B or a PC.

ACAI – Automatic Counting Accuracy Improvement
This counting function recalculates the average unit weight each time a sample
is added to eliminate errors caused by variations in unit weight.

Quick Reference Card • • • • • • • • • • • • • • • • • • • • • • • • • • • • • •A convenient operating guide is located at the bottom of the balance.

Density Measurement Function • • • • • • • • • • • • • • • • • • • •An optional Density Determination Kit (GX-13) is available.

Automatic Response Speed Adjustment (GX only)
The response rate is set automatically at either FAST, MID, or SLOW dependingon the environmental conditions. (Manual selection is available for the GF Series.)
GX-13

MONAnimal
Weighing

Comparator
Function

Auto PowerON

Auto PowerOFF

Underhook

EC TypeApproval(GX )Software

Quick USB

OptionalWeighing capacity

GX-200
GX-400 GX-600 GX-800 GX-1000
GX-2000
GX-4000 GX-6100 GX-6000 GX-8000
GF-200 GF-300 GF-400 GF-600 GF-800 GF-1000 GF-1200 GF-2000 GF-3000 GF-4000 GF-6100 GF-6000 GF-8000
210g

310g

410g

610g

810g

1100g

1210g

2100g

0.001gMinimum weighing value±0.002g±0.003g

Stabilization time (typical at FAST)

Approx. 1second

Approx. 1.5 seconds

8100g

0.1g

±0.02g±0.03gApprox. 1 second

Approx.
1.5seconds±0.1g
Approx. 1 second±5ppm/°C(auto-self calibration OFF)5°C to 40°C (41°F to 104°F), 85%RH or less (No condensation)

Operating temperature

Super Hybrid Sensor (SHS) *SHS: Patent approved

Sensing method
Display type

Vacuum Fluorescent Display (VFD)

Display refresh rate

5 times/second or 10 times/second
0.001gCounting Minimum unit massmodeNumber of samples

0.01g

0.1g

1.00g

10.0g

10, 25, 50 or 100 pieces
0.100gMinimum 100% reference mass

0.01%, 0.1%, 1% (Depends on reference mass stored)

Minimum 100% display

RS-232C Interface with Windows Communication Tools (WinCT: included as CD-ROM)

Standard serial I/F

128 x 128mm (5.04 x 5.04 inch)

165 x 165mm (6.50 x 6.50 inch)

Motor-driven Internal Calibration (GX only)

Calibration

210(W) x 317(D) x 86(H)mm (8.27 x 12.48 x 3.39 inch)

External dimensions

Confirm that the adapter type is correct for the local voltage and power receptacle

AC adapter

Approx. 11VA (supplied by AC adapter)

Power consumption

Approx. 4.6kg(GX)/3.8kg(GF)

Approx. 5.1kg(GX)/4.3kg(GF)

Notes: The internal mass is equal to (approximately) 500 grams.
EC Type Approval Confirmed (Approval No. UK2560)
Standard Breeze break included for 1mg minimum weighing value models

Dimensions WithOptional Glass Breeze Break (mm)163244

162.5451) GX-02, GX-04 GX-06 and standard RS-232C cannot be used at the same time.
2) Factory installed option

Accessories

210

AD-1682

B231.3

14186

195.5

176

Pan size (mm)ModelA

B

128

128

GX-2000/4000/6100/6000/8000
165
GF-1200/2000/3000/4000/6100/6000/8000

165

GX-200/400/600/800/1000
GF-200/300/400/600/800/1000

162.5

86

Rechargeable battery unit
(Maximum operating time: 8 hours)
AD-1683
Static eliminator
AD-1684A
Electrostatic fieldmeter
AD-1687
Weighing environment logger
AD-1688
Weighing data logger
AD-1689
Tweezers for calibration weight
AD-8127
Compact printerAD-8526
Serial / Ethernet converter
AD-8527
Quick USB adapter
AD-8920A
Remote display
AD-8922A
Remote controller
AX-USB-25P
Serial/USB converter with cableAX-KO1710-200 RS-232C cable (25P-9P)
AX-SW128
Foot switch
AX-073003691-S Dust cover for 128 x 128mm
weighing pan size models
AX-073003692-S Dust cover for 165 x 165mm
weighing pan size models

454.3

317A

GX-02 1) Quick USB Interface with cable
GX-04 1),2) Comparator Output with a Buzzer/
RS-232C/Current Loop Output
GX-06 1),2) Analog Output/Current Loop Output
GX-10 Glass Breeze Break for 128 x 128mm
weighing pan size models
GX-11 Glass Breeze Break for 165 x 165mm
weighing pan size models
GX-12 Animal Weighing Pan(not available for GX-200 and GF-200)
GX-13 Density Determination Kit for 128 x 128mm
weighing pan size models

28

Physical Dimensions (mm)317

Options

207.5

Net weight

6100g

0.1g

±2ppm/°C (When automatic self calibration is not used)

Sensitivity drift (10°C to 30°C/50°F-80°F)

Weighing pan

6100g

0.01g

Linearity

Percent
mode

4100g

0.01g

0.001gRepeatability (Standard Deviation)

3100g

210

Specifications

Specifications subject to change for improvement without notice.

AD-1687

A&D Company, Ltd.

3-23-14 Higashi-Ikebukuro, Toshima-Ku, Tokyo, 170-0013, Japan Tel: +81 3-5391-6132 Fax: +81 3-5391-1566 http://www.aandd.jp

A&D Engineering, Inc.

1756 Automation Parkway, San Jose, CA 95131, U.S.A. Tel: +1 408-263-5333 Fax: +1 408-263-0119

A&D Australasia Pty Ltd.

32 Dew Street, Thebarton, South Australia 5031, Australia Tel: +61 8-8301-8100 Fax: +61 8-8352-7409

A&D Instruments Ltd.

Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire, OX14 1DY, United Kingdom Tel: +44 1235-550420 Fax: +44 1235-550485<German Sales Office>Hamburger Straße 30, D-22926, Ahrensburg, Germany Tel: +49 4102-459230 Fax: +49 4102-459231

A&D Korea Ltd.

8F Manhattan Bldg., 33, Gukjegeumyung-ro 6-gil, Yeongdeungpo-gu, Seoul, 07331, Korea Tel: +82 2-780-4101 Fax: +82 2-782-4280

A&D Rus Co., Ltd.

Vereyskaya Str. 17, 121357, Moscow, Russia Tel: +7 495-937-33-44 Fax: +7 495-937-55-66

A&D Instruments India (P) Ltd.

509 Udyog Vihar Phase V Gurgaon-122 016, Haryana, India Tel: +91 (124) 471-5555 Fax: +91 (124) 471-5599* GX/GF-ADCC-15-PR14-18b01GP