If you take the hygiene of yourequipment seriously…
Maintaining hygienic facilities is imperative in the food-processing industry, and the sanitaryrequirements of governments and companies are becoming increasingly strict all around theworld. The SW Series of IP69K-rated Super Washdown Scales is ideally constructed forheavy-duty cleaning with high-pressure (~ 10,000 kPa) and high-temperature (~ 85 ˚C) water,
as well as for long-term use in harsh environments.

Features
IP69K - the highest level of dust andwater-proofing *Ultimate protection against the effectsof condensation
Streamlined construction for easycleaning of food-processing residues
Stainless steel weighing pan, frame,
and load cell **Rugged and adjustable indicator bracket
Withstands extremely damp
environments

Touch sensitive capacitive keys to ensure
permanent waterproofing
Bright, multicolored LED display
with 5-step comparator lamps
Auto-tare function to facilitate consecutive
check weighing
Three sizes of base units, which arealso available separately
Rechargeable battery, RS-232C + relay
output interface, and RS-422 / 485
interface (optional)

Load cell also protected at theIP69K level* IP68 compliant at the same time** Weighing pan = SUS 304
Frame = SUS 304
Load cell = SUS 630

The capacitive keys prevent
surface rupture from pressing withsharp objects and thereby ensure
permanent waterproofing.

The multicolored LED is highly
visible and intuitive.

The bracket angle is adjustable to0˚, 30˚, 60˚, and 90˚.

The pole stand can be set on thelongitudinal side.

The streamlined construction is
ideal for washdown applications.

SW-D – IP69K Indicator for High Pressure Washdown
The desktop/wall-mount standis optional.

Input sensitivity

0.2 µV / d min. (d: minimum division)

Input signal range–16 mV to 16 mV

Load cell excitation voltage

DC 5 V ± 5% (no remote sensing)Load cell drive capacity

Up to four 350

Temperature coefficient

You can purchase the SW display unit (SW-D) separately and use
it as an IP69K (as well as IP68) indicator with platforms (loadcells) other than A&D’s.

load cells

Zero±(0.2 µV ± 0.001% of zero adjustment voltage) / ˚C (typ.)Span±0.001% / ˚C of reading (typ.)Linearity±0.01% of full scale

Maximum display resolution

60000

Dimensions

255 (W) x 119 (D) x 179 (H) mm / 10.0 (W) x 4.7 (D) x 7.0 (H) inMass (approximate)

2.1 kg / 4.63 lb

Specifications
Capacitykg

Min. display

Capacitylb

Min. display

Capacityoz

Min. display

SW-6KS

SW-15KS615

SW-15KM SW-30KM SW-60KM SW-150KM SW-60KL
15306015060

SW-150KL
150

0.002

0.005

0.005

0.01

0.02

0.05

0.02

0.05

0.001 *

0.002 *

0.002 *

0.005 *

0.01 *

0.02 *

0.01 *

0.02 *

0.00050.001

0.0010.002

0.005

0.01

0.005

0.01

13

33

33

66

130

330

130

330

0.005

0.01

0.01

0.02

0.05

0.1

0.05

0.1

0.002 *

0.005 *

0.005 *

0.01 *

0.02 *

0.05 *

0.02 *

0.05 *

0.001

0.002

0.002

0.005

0.01

0.02

0.01

0.02

210

520

520

1050

2100

5200

2100

5200

0.1

0.2

0.2

0.5

12

12

0.05 *

0.1 *

0.1 *

0.2 *

0.5 *

1*

0.5 *

1*

0.02

0.05

0.05

0.1

0.2

0.5

0.2

0.5

Capacity

13

33

33

66

130

330

130

330

Min. display

0.1

0.1

0.1

0.11
1
5 pieces (Can be changed to 10, 20, 50 or 100)

1

1

Maximum count

120,000 pcs

150,000 pcs

150,000 pcs

150,000 pcs

120,000 pcs

150,000 pcs

120,000 pcs

150,000 pcs

Minimum unit weight

0.00005 kg

0.0001 kg

0.0001 kg

0.0002 kg

0.0005 kg

0.001 kg

0.0005 kg

0.001 kg

Repeatability(Std. deviation)

0.001 kg

0.002 kg

0.002 kg

0.005 kg

0.01 kg

0.02 kg

0.01 kg

0.02 kgLinearity±0.002 kg±0.005 kg±0.005 kg±0.01 kg
±0.02 kg
±0.05 kg
±20 ppm / °C (5 °C to 35 °C / 41 °F to 95 °F)±0.02 kg

±0.05 kglb-oz

Number of samples

Sensitivity drift

Weight: 7 segment LED display (Character height: 14.6 mm)
Comparator: Red/Yellow/Green/Yellow/Red LED

Display

10 times per second

Display update–10 °C to 40 °C / 14 °F to 104 °F

Operating conditions

AC main (100 V ~ 240 V) orSLA Battery (option), Approx. 90 hours (depends on usage)

Power supply

Weighing pan size

250 x 250 mm / 9.8 x 9.8 in.

300 x 380 mm / 11.8 x 15.0 in.

390 x 530 mm / 15.4 x 20.9 in.

Dimensions(for the pole type)

250(W) x 480(D) x 353.5(H) mm
9.8(W) x 18.9(D) x 13.9(H) in.

300(W) x 601(D) x 722.5(H) mm
11.8(W) x 23.7(D) x 28.4(H) in.

390(W) x 751(D) x 722.5(H) mm
15.4(W) x 29.6(D) x 28.4(H) in.

Mass (approximate)(for the pole type)

7.7 kg / 17.0 lb

13.0 kg / 28.7 lb

16.2 kg / 35.7 lb

Calibration weight(factory setting)

6 kg

15 kg

15 kg

30 kg

60 kg

150 kg

60 kg

12 lb

30 lb

30 lb

60 lb

120 lb300 lb120 lbOP-11**300 lb

*Options OP-02 Sealed lead acid rechargeable battery OP-10 Long pole standOP-03* RS-232C＋comparator relay output
OP-04* RS-422/485
OP-09 Short pole stand (for SW-6KS/15KS)

150 kgFactory setting

Desktop/wall-mount standOP-03 and OP-04 cannot be used at the same time.
The SW is not IP69K compliant with OP-03 or OP-04.

Specifications (base unit only)
SB-6K13SW SB-15K13SW SB-15K14SW SB-30K14SW SB-60K14SW SB-150K14SW SB-60K15SW SB-150K15SW
Capacity
Zero balance
Rated output

6 kg
15 kg
15 kg
30 kg
0.13 ± 0.2 mV/V
0.27 ± 0.2 mV/V 0.11 ± 0.2 mV/V
1 mV/ V ± 10%
0.5 mV/V ± 10%

60 kg

150 kg

60 kg

150 kg0.07 ± 0.2 mV/V 0.03 ± 0.2 mV/V 0.11 ± 0.2 mV/V 0.04 ± 0.2 mV/V
1 mV/V ± 10%

0.03% of rated output

Linearity & hysteresis
Compensated−10 °C to 40 °C / 14 °F to 104 °F

temprature range

DC 15 V

Max. exitation voltage
Input terminal
resistance
Output terminal
resistance

1780

± 50

1600

± 10

Greater than 5000 M

Isolation resistanceat DC 50 V

Weighing pan size

250 x 250 mm / 9.8 x 9.8 in.

ø 6 x 1.5 m, shielded cable including 4 wires
300 x 380 mm / 11.8 x 15.0 in.

Mass (approximate)

3.7 kg

8.2 kgCable

390 x 530 mm / 15.4 x 20.9 in.
11.2 kg...Clearly a Better Value
3-23-14 Higashi-lkebukuro, Toshima-ku,Tokyo 170-0013 JAPAN
Telephone:[81](3) 5391-6132 Fax:[81](3) 5391-6148
http://www.aandd.jp
1756 Automation Parkway, San Jose, CA 95131 U.S.A.
Telephone:[1](408) 263-5333 Fax:[1](408) 263-0119

A&D Australasia
ia Ptyy Ltd.
d.
32 Dew Street, Thebarton, South Australia 5031 AUSTRALIA
Telephone:[61](8) 8301-8100 Fax:[61](8) 8352-7409

Unit 24/26 Blacklands Way Abingdon Business Park,Abingdon, Oxon OX14 1DY UNITED KINGDOM
Telephone:[44](1235) 550420 Fax:[44](1235) 550485

Vereyskaya str.17, Moscow, 121357 RUSSIA
Telephone: [7] (495) 937-33-44 Fax: [7] (495) 937-55-66<German Sales Office>509 Udyog Vihar Phase V
Gurgaon-122 016, Haryana, INDIA
Telephone: [91](124) 471-5555 Fax: [91](124) 471-5599

Große Straße 13 b 22926 Ahrensburg GERMANY
Telephone:[49](0) 4102 459230 Fax:[49](0) 4102 459231

A&D
D Instruments
Inst uments India Private
Pre Limited
mited

Manhattan Bldg. 8F, 36-2 Yoido-dong, Youngdeungpo-gu, Seoul, KOREA
Telephone:[82](2) 780-4101 Fax:[82](2) 782-4280*SW-ADCC-02-PR2-12403