Product Information

IBV 100 Series
Interpolation andDigitizing Electronics

May 2010

IBV 100 Series
Interpolation and digitizing electronics• Input signals » 1 VPP• Output signals « TTLTolerancing ISO 8015
ISO 2768 - m H< 6 mm: ±0.2 mm

98

¬ 4.5

14.5

6147
173

A

 0.3 A

À

163±0.2

 0.3 A86±0.252±0.2

25

M4

¬ 4.5

R55

1362

Adapter plate: ID 536 452-01

7±1

¬ 7.5

1Accessories:
An adapter plate is available for mounting
on existing holes of the IBV 6xx/EXE 6xx:64M4*)36±0.2

2 mounting screwsM4 x 16 DIN 912/ISO 4762

21.5*)

86±0.236±0.238.5±186±0.2

 0.3 A

M4¬ 4.5
Adapter plate

M4
Md = 2.5 Nm
Included in items supplied.

M4

Product Information for IBV 100 Series

5/2010Specifications

IBV 101
IBV 102Input» 1 VPPElectrical connection

M23 flange socket (female) 12-pin

Cable length† 60 m for UP > 4.9 V† 30 m at IEncoder † 120 mA1)

Interpolation

IBV 101: 5-fold, 10-fold
IBV 102: 20-fold, 25-fold, 50-fold, 100-fold

Input frequency1) forinterpolation

Nominal values2)

IBV 101

IBV 102

5-fold 200 kHz

200 kHz133 kHz100 kHz80 kHz

50 kHz

25 kHz10-fold 200 kHz

100 kHz66 kHz

50 kHz40 kHz

25 kHz12.5 kHz

20-fold 100 kHz

50 kHz33 kHz

25 kHz

20 kHz12.5 kHz6.25 kHz

25-fold 80 kHz

40 kHz

26 kHz

20 kHz

16 kHz

10 kHz

5 kHz

50-fold 40 kHz

20 kHz

13 kHz

10 kHz

8 kHz

5 kHz

2.5 kHz

100-fold 20 kHz

10 kHz6.6 kHz

5 kHz

4 kHz

2.5 kHz1.25 kHz‡ 0.950 µs‡ 1.925 µs

Output1)« TTL (clocked)

Electrical connection

M23 flange socket (male) 12-pin

Cable length† 25 m† 50 m† 75 m(¥ † 50 m)† 100 m (¥ † 50 m)

Edge separation a‡ 0.100 µs‡ 0.220 µs‡ 0.345 µs‡ 0.465 µs

Reference mark signal1)

Pulse width 90° elec. or 270° elec.

Fault indication1)

through fault detection signal ¥ or, in addition, Ua1/Ua2 high impedancePower supply

5V ± 5 %3)

Current consumption

IBV 101: † 120 mA
IBV 102: † 130 mA

Operating temperature
Storage temperature

0 °C to 70 °C–30 °C to 80 °C

Vibration 50 to 2 000 Hz
Shock 11 ms† 100 m/s2
† 300 m/s2

Degree of protection

IP 65

WeightApprox. 0.3 kg‡ 0.585 µs

Bold: This version is the factory default setting, please select when ordering
Adjustable
2)
The actual input frequency can be up to 5% lower. Exceeding this limit results in failure3)
Not including output load (80 mA with recommended input circuitry) or the current consumption of the encoder(see the corresponding encoder brochure)1)

Product Information for IBV 100 Series

5/2010Electrical Connection

Connecting cable oradapter cable with
M23 connector (male) 12-pin

M23 connecting cable12-pin, ¬ 8 mmComplete
ID 298 399-xx«»

Cable and connector, 12-pin.
See also HEIDENHAIN catalogs
for linear encoders, angle
encoders and rotary encoders
as well as Product Information
sheets for the respectiveencoders

With one connector
ID 309 777-xx

22[4(2x0.14mm ) + (4x0.5 mm )]Cable only
ID 244 957-01
Connector (female), 12-pin
ID 291 697-05

IBV input – » 1 VPP12-pin M23 flange socket

91
2108
121137654

Power supply

Incremental signals

Other signals

122101156813479UPSensor
UP0VSensor
0VA+A–

B+B–

R+R–

Vacant Vacant Vacant

Brown/
GreenBlueWhite/
GreenWhiteBrown

GreenGray

Pink

RedBlack

Violet//Yellow

Shield on housing; UP = Power supply voltage
Sensor: The sensor line is connected internally with the corresponding power line.

IBV output – « TTL
12-pin M23 flange socket

12-pin M23 connector

Power supply

Incremental signals

122

10

115

UP

Sensor5V

0V

Sensor0V

Ua1Brown/
GreenBlueWhite/
GreenWhite

Brown

6GreenOther signals

813479Ua2£

Ua0¤¥PWT testpin

Gray

PinkRedBlack

VioletYellow

Cable shield connected to housing; UP = power supply voltage
Sensor: The sensor line is connected in the encoder with the corresponding power line

For more information• Product overview: Interface Electronics
DR. JOHANNES HEIDENHAIN GmbH
Dr.-Johannes-Heidenhain-Straße 583301 Traunreut, Germany{ +49 8669 31-0
| +49 8669 5061
E-mail: info@heidenhain.dewww.heidenhain.de624 603 · 01 · A · 02 · 5/2010 · PDF