FG-CWP/FG-ACWP Series
Quick Start Guide
FG-CWP/FG-ACWP Series
For Water proof platform scale

English (EN)Guide de démarrage rapide
Séries FG-CWP/FG-ACWP
Balance numérique à plateforme
numérique étanche

Français (FR)

Schnellstartanleitung
Serie FG-CWP/FG-ACWP
Wasserdichten digitalen Plattformwaage

Deutsch (DE)

Guía de inicio rápido
Series FG-CWP/FG-ACWP
Báscula de plataforma digital
impermeable

Español (ES)Guida introduttiva
Serie FG-CWP/FG-ACWP
Bilancia digitale impermeabile
a piattaforma

Italiano (IT)

快速入門指南FG-CWP/FG-ACWP 系列
防水數位平台秤

中文 (CT)

1WMPD4004365C

Warning definition
The warning definition described in this manual has the following meaning:
DANGERAN imminently hazardous situation which, if not
avoided, will result in death or serious injury.

© 2022 A&D Company, Limited. All rights reserved.
No part of this publication may be reproduced, transmitted, transcribed, or translated
into any language in any form by any means without the written permission of A&D
Company,Limited.

The contents of this publication and the specifications of the instrument covered by thispublication are subject to change for improvement without notice.

Quick Start Guide – FG-CWP/FG-ACWP Series

EN

Water proof digital platform scale……………………………….4

Guide de démarrage rapide – Séries FG-CWP/FG-ACWP

FRBalance numérique à plateforme numérique étanche…… 11

Schnellstartanleitung – Serie FG-CWP/FG-ACWP

DE

Wasserdichten digitalen Plattformwaage……………………18

Guía de inicio rápido – Series FG-CWP/FG-ACWP

ES

Báscula de plataforma digital impermeable….………………25

Guida introduttiva – Serie FG-CWP/FG-ACWP

IT

Bilancia digitale impermeabile a piattaforma………….……..32

快速入門指南– FG-CWP/FG-ACWP 系列

CT

防水數位平台秤………………………………………………..…39

EN

1. Introduction
This quick start guide describes the basic functions and operations of the scale.
The functions and operating procedures of the scale (FG-CWP/FG-ACWP series) are explained
in details in the following instruction manual.
[FG-CWP/FG-ACWP Series Instruction Manual]This is an instruction manual for understanding and fully utilizing the water proof digital platform scale FGCWP/FG-ACWP series. This manual can be downloaded from our website https://www.aandd.jp.
Scan this QR code for this manual.

2. Cautions
2.1. Precautions for Installing the Scale
DANGER□ Ground the scale, so that the user will not be subjected to an electric shock.
□ To prevent electric shock, do not handle the USB cable with wet hands.
□ The USB cable is not water resistance. Use a USB outlet located at a place where the plug
will not get wet. (For FG-CWP series (USB power model))□ The AC plug is not water-resistant. Use an electrical outlet located at a place where the plug
will not get wet. (For FG- ACWP series (AC power model))□ Do not install the scale where flammable or corrosive gas is present.
□ Do not put excessive force on cables.
□ The scale is heavy. Use caution, as lifting may cause it to fall over.
□ Use shielded load cell cables.

Consider the following installation conditions to get the greatest performancefrom your scale.
□ The ideal conditions for installation are stable temperature and humidity, solid and level surface,location with no draft or vibration, indoors out of direct sunlight and a stable power supply.
□ Do not install on a soft floor or where there is vibration.
□ Do not install where there is wind or large fluctuations in temperature.
□ Avoid locations in direct sunlight.
□ Do not install in locations with strong magnetic fields or strong radio signals.
□ Do not install where static electricity is likely to occur. When humidity is 45% R.H. or less,plastic and insulating materials are apt to be charged with static electricity due to friction, etc.
□ Unstable AC power supply may cause malfunctions.
□ Remove the protective film from the weighing pan before use.
□ Sensitivity adjustment should be performed for more accurate weighing when the scale isfirst installed or when it is moved far. Refer to the FG-CWP/FG-ACWP Series Instruction
Manual on our website https://www.aandd.jp.

2.2. Precautions for Storing□ Do not disassemble the scale.
□ If the scale becomes dirty, wipe it with a soft cloth slightly moistened with a mild detergent or
a rubbing alcohol (70% or less). Do not use organic solvents.
□ Do not scrub with a brush or the like.
□ Do not spray with strong jets of water.
4

EN

2.3. Precautions for Weighing□ Do not place loads on the weighing pan that exceed the weighing capacity.
□ Periodic sensitivity adjustment is recommended in order to maintain accurate weighing.
□ Do not apply shock to or drop anything on the weighing pan.
□ Do not use a sharp instrument such as a pencil or pen to press keys or switches.
□ Press the ZERO key before each weighing to reduce the chance of error.
□ Do not weigh with the scale immersed in water.
□ Periodically confirm that weighing values are correct.
□ The panel on the rear of the display unit should be kept closed when using the scale inorder to make it dustproof and waterproof.

3. Items Included
The following items are packed.
<FG-CWP series (USB power model)>● Main Unit
FG-30KCMWP● Quick Start Guide
FG-60KCLWP
FG-150KCLWP● USB cable(Approx. 3m)● AC Adapter(Image diagram)<FG-ACWP series (AC power model)>● Main Unit
FG-30KACMWP● Quick Start Guide
FG-60KACLWP
FG-150KACLWP● AC cable(Approx. 3m)

5EN

4. Part Names<FG-CWP series (USB power model)>Display unit (SUS304)
Rating Label
Serial Number

USB Cable（Approx. 3m）
USB Plug

Pole（SUS304）

Weighing Pan（SUS304）

Load Cell
Cable

Base Unit（SUS304）

Knob for Angle
Adjustment

BubbleSprit Level

Rear of Display Unit
OK

Leveling FeetNG

<FG-ACWP series (AC power model)>Rating Label
Serial Number

Knob for Angle
Adjustment

Display unit (SUS304)
AC Cable（Approx. 3m）
Pole（SUS304）

AC Plug

AC Cable（Approx. 3m）

Weighing Pan（SUS304）

Load Cell
Cable

Base Unit（SUS304）

Load Cell Panel
BubbleSprit Level

Earth
Terminal
Earth

Rear of Display UnitOK

NG

Leveling Feet

4.1. Display and Symbols
Display Unit

Weighing
Condition
Comparator LED

Conditions for
Each Function

Weighing Units

6

EN

Display/symbol

Description

STABLE 〇This is lit when the weighing value is stable, indicating that the scale is in
the proper condition for reading weighing values.

NET

This is lit when weight of the container (tare) is subtracted.

ZERO

This is lit when the scale is at the zero point (reference point for weighing).

Weighing Units“kg”, “g”, “pcs”, “oz” and “lb” is lit.
Note: “g” is only used with the FG-30KCMWP/FG-30KACMWP.

This is lit when the display is held.
This is a function to detect impact to the mass sensor section and to
display the impact level.
Lights up when successfully connected to the optional FG-27CWP(Bluetooth communication interface).
While the comparator function is being used, the weighing value iscompared using the preset threshold values and the indicator displays theresult.
Alternating current.
Earth (ground) Terminal.

□ For details on the method of operation and various functions, refer to the FG-CWP/FG-ACWP
Series Instruction Manual on our website https://www.aandd.jp.
□ However, please note that functions related to USB and mobile battery cannot be used with
FG-ACWP Series (AC power model).

4.2. Operation Keys
Operation key

Description
ON/OFF Key
Shows or hides the display.
Note: When the power plug of the scale is connected, power is supplied tothe electric circuits inside the scale.
SAMPLE Key
Selecting “pcs” as the unit activates individual weight setting mode.
Pressing and holding (for about 2 seconds) activates the comparator
threshold setting mode
PRINT Key
Outputs the weighed value as data.
Increases the value of the digit flashing when setting.
MODE Key
Switches the weighing unit.
Shifts to the right the digit flashing when making settings.
ZERO Key
Based on the zero point weighed at the time of turning on the display, bypressing the ZERO key when the weight value is stable within ±2% of theweighing capacity, the zero point indicator turns on as soon as the display
becomes zero.

7

EN

TARE KeyThe weight of the item on the weighing pan is subtracted as the weight ofa container (tare weight).

5. Setup
1.

Take the base unit and the pole out of thepacking box while being careful not to pull theload cell cable.

Display Unit

2.
2. Remove the weighing pan from the base.

3.

Assembling the pole
Temporarily remove the screws attached to thebottom of the weighing base, and then fasten
the pole and base with the removed screws,
taking care that the load cell cable is not pinched.
Bundle and insert the excess load cell cable into
the pole from the display side.

Pole
Weighing
Pan

1.
Base Unit
Load Cell Cable

4. Put the weighing pan on the base unit.

3.
If using the scale in a location where static electricity iseasily generated or where it may get wet, secure thegrounding wire with the earth (ground) terminal (M4screw).
Earth(Ground)
Terminal
Rear of Display Unit

6.

Maintenance・Refer to the FG-CWP/FG-ACWP Series Instruction Manual on our website https://www.aandd.jp
for sensitivity adjustment.
・Periodically check the accuracy of weighing. Adjust the scale, if it is moved to another location or theenvironment has changed.

6.1.

RepairDo not disassemble / assemble the scale without an authorized service engineer. Doing so may
cause an electric shock or damage to the scale, etc. In this case, repair is not covered underwarranty. Contact your local A&D dealer if your scale needs service or repair.

8

EN

6.2.

Error display

In this situation

Confirm these items
・Is the main power the correct voltage?
・Is the main power cord properly connected?
・Check around the weighing pan.
・Is there anything on the weighing pan?
・Perform zero point sensitivity adjustment.

Nothing displayed.
Scale does not turn on.
The scale does not
display zero at first.

・Weighing error that meaning "Over loaded."is displayed.

is displayed.

・Indicates that the sample weight is too light to set the unit weight inthe counting mode.
・Sensitivity adjustment error that meaning "Too light."

is displayed.

・Displayed when the main power drops.

is displayed.

・The weighed value is unstable due to drift, vibration or other factors.
A breeze or vibration may be affecting measurement.
Check around the weighing pan.
・Check the connection of the load cell cable.
・Did you use the "hold function" ?
Change the function.
・Turn off scale and turn it on again.

is displayedand does not proceed.
Fixed display.

6.3.

Error code table

When any of the following errors are displayed, try turning the display off and on again.

Display

Content

is displayed

The temperature sensor has failed.

is displayed

The memory (circuit) has failed.

is displayed

The internal circuitry has failed.

is displayed

The mass sensor has failed.

7. Specifications
Models
Weighing Capacity

kg

Readability
Weighing Capacityg

Readability
Weighing Capacitylb

Readability

FG-30KCMWP
FG-30KACMWP
30
0.01
0.005 *
0.002
0.001
30000
10
5*
2
1
66
0.02
0.01 *
0.005
0.002
9

FG-60KCLWP
FG-60KACLWP
60
0.02
0.01 *
0.005
0.002

FG-150KCLWP
FG-150KACLWP
150
0.05
0.02 *
0.01
0.005

130
0.05
0.02 *
0.01
0.005

330
0.1
0.05 *
0.02
0.01

EN

Models
Weighing Capacity

oz

lb-oz

Readability
Weighing Capacity
Readability

Number of Samples
Maximum Count Number
Minimum Unit Mass [kg]Repeatability(Standard Deviation) [kg]Linearity [kg]Stabilization time
(good environment)
Temperature Drift
Display
Display update
Dust and water protection
Operating Conditions
Installation environment
AltitudePower Supply

Overvoltage category **Pollution degree **Weighing Pan Size [mm]Dimension [mm]Width x Depth x Height
Weight (approximately)

FG-30KCMWP
FG-60KCLWP
FG-150KCLWP
FG-30KACMWP
FG-60KACLWP
FG-150KACLWP
1050
2,100
5,200
0.5
1
2
0.2 *
0.5 *
1*
0.1
0.2
0.5
0.05
0.1
0.2
66
130
330
0.1
1
1
5 pcs (can be changed to 10 pcs , 20 pcs, 50 pcs or 100 pcs)
300,000 pcs
0.00010.00020.0005
0.005

0.01

0.02

±0.01

±0.02
±0.05
20kg:Approx. 1.0 sec (Factory setting)Approx. 0.8 sec (setting)±20 ppm/°C typ. (5 °C to 35 °C)
7 segment LCD with backlight(Character height 30 mm)
Approx. 10 times/second
Complies with IP67
-10 °C to 40 °C, 85% R.H. or less (non-condensing)Indoor use only
Altitude up to 2,000 m<FG-CWP Series (USB power model)>Power supplied from AC adapter, mobile battery, or USB Type-A
port
USB Cable Length: Approx. 3 m(USB cable is also used for communication.)<FG-ACWP Series (AC power model)>AC mains (100 V to 240 V, 50/60 Hz, 0.1 A Max)
AC Cable Length: Approx. 3 m
II
2
300 x 380
370 x 500
300 x 515 x 784

370 x 635 x 784

11.1 kg / 24.5 lb

12.8 kg
28.3 lb

Sensitivity adjustment
30 kg / 60 lb
60 kg / 120 lb
150 kg / 300 lb
weight (factory setting)* Factory setting** Applicable to FG-ACWP Series (AC power model) only
FG-CWP/FG-ACWP Series Class Ⅲ Model Specifications
FG-30KCMWP
FG-60KCLWP
FG-150KCLWP
Models
FG-30KACMWP
FG-60KACLWP
FG-150KACLWP
Accuracy class
Ⅲ
Maximum Capacity
30
60
150
kg
Readability
0.01
0.02
0.05
Maximum Capacity
66
130
330
lb
Readability
0.02
0.05
0.1
Maximum Capacity
1,050
2,100
5,200
oz
Readability
0.5
1
2
0.2
0.4
1
Minimum Unit Mass [kg]
30
60
150
Maximum tare [kg]□ Class Ⅲ approved model specifications. For details, refer to the FG-CWP/FG-ACWP Series
Instruction Manual on our website https://www.aandd.jp. The appearance is the same as thatof the FG-CWP/FG-ACWP series.
10FR

1. Introduction
Ce guide de démarrage rapide décrit les fonctions et les opérations de base de la échelle.
Les fonctions et les procédures de fonctionnement de la balance (série FG-CWP/FG-ACWP) sontexpliquées en détail dans le manuel d'instructions suivant.
[Manuel d'instructions de la série FG-CWP/FG-ACWP] Il s'agit d'un manuel d'instructions
permettant de comprendre et d'utiliser pleinement la balance à plate-forme numérique étanchesérie FG-CWP/FG-ACWP. Ce manuel peut être téléchargé depuis notre site Web
https://www.aandd.jp.
Scannez ce code QR pour ce manuel.

2. Précautions
2.1. Précautions d'installation de la balance
DANGER□ Mettre la balance à la terre afin de protéger l'utilisateur des chocs électriques.
□ Afin d'éviter les chocs électriques, ne pas manipuler le câble USB avec les mains mouillées.
□ Le câble USB n'est pas résistant à l'eau. Utilisez une prise USB située à un endroit où la fiche
ne sera pas mouillée (pour la série FG-CWP (modèle alimenté par USB)).
□ La fiche CA n'est pas résistante à l'eau. Utilisez une prise électrique située à un endroit où lafiche ne sera pas mouillée (pour la série FG-ACWP (modèle alimenté par CA)).
□ Ne pas installer la balance dans un environnement exposé à des gaz inflammables ou corrosifs.
□ Ne pas soumettre les câbles à une force excessive.
□ La balance est lourde. Faire preuve de prudence lorsque la machine est soulevée, car ellerisque de basculer.
□ Utilisez des câbles de cellule de pesée blindés.

Tenir compte des conditions d'installation suivantes pour obtenir les
meilleures performances de la balance.
□ Dans l'idéal, installer la balance dans un endroit où la température et l'humidité sont stables,
sur une surface plane et solide, sans vibrations ni risque de décalage, en intérieur, à l'abri dela lumière directe du soleil et avec une alimentation stable.
□ Ne pas installer sur un sol mou ou soumis à des vibrations.
□ Ne pas installer dans un endroit exposé au vent ou à d'importants changements de température.
□ Éviter les endroits exposés à la lumière directe du soleil.
□ Ne pas installer dans des endroits exposés à de puissants champs magnétiques ou signaux radio.
□ Ne pas installer dans des endroits susceptibles d'être exposés à de l'électricité statique.
Lorsque l'humidité est de 45 % HR ou moins, les matériaux plastique et d'isolation peuvent se
charger d'électricité statique en raison de la friction, etc.
□ Une alimentation CA instable risque de provoquer des dysfonctionnements.
□ Retirer le film de protection du plateau de la balance avant l'utilisation.
□ Régler la sensibilité pour un pesage plus précis lors de l'installation initiale de la balance ou
après un long transport. Reportez-vous au manuel d'instructions des séries FG-CWP/FGACWP sur notre site Web https://www.aandd.jp.

2.2. Précautions de stockage□ Ne pas démonter la balance.
□ Si la balance est sale, l'essuyer avec un chiffon doux légèrement humidifié avec un détergent
doux ou un alcool à friction (70% ou moins). Ne pas utiliser de solvants organiques.
□ Ne pas frotter avec une brosse ou objet similaire.
□ Ne pas projeter de puissants jets d'eau.

11

FR

2.3. Précautions de pesage□ Sur le plateau de la balance, placer uniquement des charges inférieures à la capacité depesage.
□ Il est recommandé de régler régulièrement la sensibilité pour assurer la précision du pesage.
□ Protéger le plateau de la balance des chocs et ne rien faire tomber dessus.
□ Ne pas utiliser d'objet pointu, par exemple un crayon ou un stylo, pour appuyer sur les
touches ou les commutateurs.
□ Appuyer sur la touche ZERO avant chaque pesage pour réduire le risque d'erreur.
□ Ne pas utiliser la balance si elle est en contact avec de l'eau.
□ Vérifier régulièrement que les valeurs de pesage sont correctes.
□ Le panneau situé à l'arrière de l'unité d'affichage doit rester fermé lorsque la balancefonctionne afin d'assurer l'étanchéité à la poussière et à l'eau.

3. Articles inclus
Le produit est emballé avec les articles suivants.
< Série FG-CWP (modèle alimenté par USB)>● Unité principale
FG-30KCMWP● Guide de démarrage rapide
FG-60KCLWP
FG-150KCLWP● CâbleUSB(environ 3 m)

● Adaptateur CA(Schéma illustratif)< Série FG-ACWP (modèle à alimentation CA>● Unité principale
FG-30KACMWP● Guide de démarrage rapide
FG-60KACLWP
FG-150KACLWP● CâbleCA(environ 3 m)

12

FR

4. Nom des pièces< Série FG-CWP (modèle alimenté par USB)>
Numéro desérie del'étiquette declassement

Unité d'affichage(SUS304)
Câble USB(environ 3 m)

Montant(SUS304)

Prise USB
Plateaude la balance(SUS304)

Câble de lacellule de charge
Niveau à
bulle

OK

Socle
(SUS304)

Molette deréglage del'angleArrière del'unité d'affichage

NG

Pieds réglables

Numéro desérie del'étiquette declassement< Série FG-ACWP (modèle alimenté par CA)>
Molette de réglage
de l'angle

Unité d'affichage (SUS304)
Câble CA（environ. 3m）
Montant（SUS304）

Prise CA
Plateaude la balance(SUS304)

Socle（SUS304）

Câble CA
（environ. 3m）

Câble de lapanneau de cellule
cellule de charge
de charge
Niveau à
Terre
bulle

OK

NG

Borne deterre

Arrière del'unité d'affichage

Pieds réglables

4.1. Affichage et symboles
Unité d'affichage

Conditionde pesage
LED de comparateur

Conditions dechaque fonction

Unités de pesage

13

FR

Affichage/symbole

Description

STABLE 〇Ceci s'allume lorsque la valeur de pesage est stable afin d'indiquer que labalance dispose des bonnes conditions pour lire les valeurs de pesage.

NET
ZERO
Unités de pesage

Ceci s'allume quand le poids du contenant (tare) est soustrait.
Ceci s'allume quand la balance se trouve au point zéro (point deréférence pour le pesage).
« kg », « g », « pcs », « oz » et « lb » s'allument.
Remarque : l'unité « g » n'est utilisée qu'avec le modèle FG-30KCMWP.
/FG-30KACMWP.
Ceci s'allume lorsque l'écran est tenu.
Cette fonction sert à détecter l'impact sur la section du capteur demasse et à afficher le niveau d'impact.
S'allume lorsque la connexion au FG-27CWP en option est établie(interface de communication Bluetooth).
Lorsque la fonction de comparateur est utilisée, la valeur de pesage est
comparée à l'aide des valeurs de seuil prédéfinies et l'indicateur affichele résultat.
Courant alternatif.
Borne de terre (masse).

□ Pour plus de détails sur la méthode d'utilisation et les diverses fonctions, reportez-vous au manueld'instructions des séries FG-CWP/FG-ACWP sur notre site Web https://www.aandd.jp.
□ Cependant, veuillez noter que les fonctions liées à l'USB et à la batterie mobile ne peuventpas être utilisées avec le Série FG- ACWP (modèle alimenté par secteur).

4.2. Touches de commandeTouche decommande

Description
Touche ON/OFF
Affiche ou masque l'affichage.
Remarque : lorsque la fiche d'alimentation de la balance est connectée,les circuits électriques à l'intérieur de la balance sont alimentés.
Touche SAMPLE
Lors de la sélection de « pcs » comme unité, le mode de réglage individuel
du poids est activé.
Une pression longue (d'environ 2 secondes) active le mode de réglage duseuil du comparateur.
Touche PRINT
Fournit la valeur de pesage sous forme de données.
Augmente la valeur du chiffre qui clignote lors du réglage.
Touche MODE
Change l'unité de pesage.
Déplace vers la droite le chiffre qui clignote lors des réglages.
Touche ZERO
D'après le point zéro pesé lors de l'activation de l'affichage, lors d'unepression sur la touche ZERO quand la valeur du poids est stable à ±2 %de la capacité de pesage, l'indicateur du point zéro s'allume dès quel'affichage passe à zéro.

14

FR

Touche TARE
Le poids de l'article posé sur le plateau de la balance est soustrait en tantque poids de contenant (tare).

5. Configuration
1.

Sortir le socle et le montant de l'emballage enveillant à ne pas tirer sur le câble de la cellulede charge.

Unité d'affichage

2.

3.

Montant

Plateau dela balance

2. Retirer le plateau du socle.

Assemblage du montant
Retirer temporairement les vis fixées sur ledessous du socle de pesage, puis attacher lemontant et le socle avec les vis retirées, enveillant à ne pas pincer le câble de la cellule decharge. Regrouper et insérer l'excédent decâble de la cellule de charge dans le montant
depuis le côté affichage.

1.
Socle
Câble de lacellule de charge

4. Placer le plateau de la balance sur le socle.

3.
Si vous utilisez la balance dans un endroit oùl'électricité statique est facilement générée ou où ellepeut être mouillée, fixez le fil de terre avec la bornede terre (masse) (vis M4).
Borne protectricede mise à la terre

Arrière de l'unité d'affichage

6.

Maintenance・Reportez-vous au manuel d'instructions des séries FG-CWP/FG-ACWP sur notre site Web
https://www.aandd.jp pour le réglage de la sensibilité.
・Vérifier régulièrement la précision du pesage. Régler la balance si elle a été déplacée dans un
nouvel endroit ou si l'environnement a changé.

6.1.

Réparation

Ne démonter/monter la balance qu'en présence d'un technicien de service agréé. Sinon, l'utilisateur risquede subir un choc électrique ou d'endommager la balance, etc. Dans ce cas, la garantie ne couvre pas laréparation. Contacter le distributeur A&D local si la balance a besoin d'un entretien ou d'une réparation.

15

FR

6.2.

Affichage des erreurs

Dans cette situation
Vérifier ces éléments
Rien ne s'affiche.
・L'alimentation principale est-elle à la bonne tension ?
La balance ne démarre・Le cordon de l'alimentation principale est-il correctement branché ?
pas.
・Vérifier les abords du plateau de la balance.
La balance n'affiche・Y a-t-il quelque chose sur le plateau de la balance ?
pas le zéro en premier.
・Ajustez la sensibilité du point zéro.
・Erreur de pesage signifiant « Surcharge ».
s'affiche.
s'affiche.

・Indique que le poids d'échantillon est trop faible pour régler le poids
de l'unité en mode de comptage.

s'affiche.

・Erreur de réglage de la sensibilité signifiant « Trop léger ».

s'affiche.

・S'affiche lorsque l'alimentation principale chute.
・La valeur pesée est instable en raison de la dérive, des vibrations
ou d'autres facteurs.
Un courant d’air ou une vibration peut affecter la mesure.
Vérifier autour du plateau de la balance.
・Vérifier la connexion du câble de la cellule de charge.
・La « fonction de maintien » a-t-elle été utilisée ?
changer la fonction・Arrêter la balance, puis la redémarrer.

s'affiche etne poursuit pas.

Affichage fixe.

6.3.

Tableau des codes d'erreur

Lorsque l'une des erreurs suivantes s'affiche, arrêter l'écran et le redémarrer.

Affichage

Contenus'affiche

Le capteur de température a dysfonctionné.

s'affiche

La mémoire (circuit) a dysfonctionné.

s'affiche

Les circuits internes ont dysfonctionné.

s'affiche

Le capteur de masse a dysfonctionné.

7. Spécificités
ModèlesCapacité de pesage

kg

Lisibilité
Capacité de pesageg

Lisibilité
Capacité de pesagelb

Lisibilité

FG-30KCMWP
FG-30KACMWP
30
0,010,005*
0,002
0,001
30 000
10
5*
2
1
66
0,02
0,01*
0,005
0,002
16

FG-60KCLWP
FG-60KACLWP
60
0,02
0,01*
0,005
0,002

FG-150KCLWP
FG-150KACLWP
150
0,05
0,02*
0,01
0,005

130
0,05
0,02*
0,01
0,005

330
0,1
0,05*
0,02
0,01

FR

ModèlesCapacité de pesage

oz

lb-oz

Lisibilité
Capacité de pesage
Lisibilité

Nombre d'échantillons
Nombre maximal pouvantêtre compté
Masse unitaire minimale [kg]Répétabilité(Écart type) [kg]Linéarité [kg]Temps de stabilisation(bon environnement)
Dérive thermiqueAffichage
Actualisation de l'affichage
Protection contre lapoussière et l'eau
Conditions d'exploitation
Environnement d'installation
Altitude

Alimentation

Catégorie de surtension **Degré de pollution **Dimensions du plateau dela balance [mm]Dimensions [mm]Largeur x Profondeur x
HauteurPoids (approximatif)

FG-30KCMWP
FG-60KCLWP
FG-150KCLWP
FG-30KACMWP
FG-60KACLWP
FG-150KACLWP
1 050
2 100
5 200
0,5
1
2
0,2*
0,5*
1*
0,1
0,2
0,5
0,05
0,1
0,2
66
130
330
0,1
1
1
5 pcs (possibilité de modifier sur 10 pcs, 20 pcs, 50 pcs ou 100 pcs)
300 000 pcs
0,00010,005±0,01

0,00020,0005
0,01
0,02
±0,02
±0,05
20kg:Approx. 1.0 sec (Factory setting)Approx. 0.8 sec (setting)±20 ppm/°C typ. (5 °C à 35 °C)
LCD 7 segments avec rétroéclairage (hauteur des caractères : 30 mm)
Environ 10 fois/secondeConforme à la norme IP67
-10 °C à 40 °C, 85 % HR ou moins (sans condensation)
Utilisation en intérieur uniquement
Altitude jusqu'à 2 000 m<Série FG-CWP (modèle alimenté par USB)>
Adaptateur CA, batterie mobile ou port USB
Longueur du câble USB : environ 3 m(Le câble USB est également utilisé pour la communication.)<Série FG-ACWP (modèle alimenté par CA>
Secteur CA (100 V à 240 V, 50/60 Hz, 0.1 A Max)
Longueur du câble CA: environ. 3 m
II
2
300 x 380

370 x 500

300 x 515 x 784

370 x 635 x 784

11,1 kg / 24,5 lb

12,8 kg / 28,3 lb

Poids du réglage de la
30 kg / 60 lb
60 kg / 120 lb
150 kg / 300 lb
sensibilité (réglage d'usine)*
Réglage d'usine** Applicable uniquement au Série FG-ACWP (modèle à alimentation CA)
Spécificités du modèle de classe Ⅲ de la série FG-CWP/FG-ACWP
FG-60KCLWP
FG-150KCLWP
FG-30KCMWP
Models
FG-30KACLWP
FG-60KACLWP
FG-150KACLWPⅢClasse de précision
Capacité maximale
30
60
150
kg
Lisibilité
0.01
0.02
0.05
Capacité maximale
66
130
330
lb
Lisibilité
0.02
0.05
0.1
Capacité maximale
1,050
2,100
5,200
oz
Lisibilité
0.5
1
2
0.2
0.4
1
Masse unitaire minimale [kg]
30
60
150
Tare maximale [kg]□ Spécifications du modèle approuvé de classe Ⅲ. Pour plus de détails, reportez-vous aumanuel d'instructions des séries FG-CWP/FG-ACWP sur notre site Web
https://www.aandd.jp.. L'apparence est la même que celle de la série FG-CWP/FG-ACWP.
17

DE

1. Einführung
In dieser Schnellstartanleitung werden die Grundfunktionen und die Bedienung der Skala beschrieben.
Die Funktionen und die Bedienungsverfahren der Waage (Serie FG-CWP/FG-ACWP) werden indem folgenden Bedienungshandbuch näher erklärt.
[Bedienungshandbuch der Serie FG-CWP/FG-ACWP] Dies ist eine Bedienungsanleitung zum
Verständnis und zur vollständigen Nutzung der wasserdichten digitalen Plattformwaagen der FGCWP/FG-ACWP-Serie. Dieses Handbuch kann von unserer Website https://www.aandd.jp.
heruntergeladen werden.
Scannen Sie diesen QR-Code für dieses Handbuch.

2. Vorsichtshinweise
2.1. Vorsichtsmaßnahmen zum Installieren der Waage
GEFAHR□ Erden Sie die Waage, damit der Benutzer keinen Stromschlag erleidet.
□ Um Stromschläge zu vermeiden, fassen Sie das USB-Kabel nicht mit feuchten bzw. nassen Händen an.
□ Das USB-Kabel ist nicht wasserdicht. Verwenden Sie eine USB-Steckdose an einer Stelle, ander der Stecker nicht nass wird. (Für FG-CWP-Serie (USB-Stromversorgungsmodell))□ Der Netzstecker ist nicht wasserdicht. Verwenden Sie eine Steckdose an einem Ort, an dem
der Stecker nicht nass wird. (Für FG-ACWP-Serie (AC-Modell))□ Stellen Sie die Waage nicht in einer Umgebung auf, in der entzündliche oder ätzende Gase
auftreten.
□ Üben Sie keine übermäßige Kraft auf die Kabel aus.
□ Die Waage ist schwer. Seien Sie vorsichtig, da sie beim Hochheben umfallen kann.
□ Verwenden Sie abgeschirmte Wägezellenkabel.

Beachten Sie die folgenden Installationsbedingungen, um die bestmögliche
Leistung Ihrer Waage zu erzielen.

□ Die idealen Installationsbedingungen sind eine stabile Temperatur und Luftfeuchtigkeit, eine
feste und ebene Oberfläche, ein Standort ohne Zugluft oder Vibrationen, ein Innenraum ohne
direkte Sonneneinstrahlung und eine stabile Stromversorgung.
□ Stellen Sie das Gerät nicht auf einem weichen Boden oder an einem Ort auf, an dem es
Vibrationen ausgesetzt ist.
□ Stellen Sie das Gerät nicht an einem Ort auf, an dem es Wind oder starken
Temperaturschwankungen ausgesetzt ist.
□ Vermeiden Sie Standorte mit direkter Sonneneinstrahlung.
□ Installieren Sie das Gerät nicht an Orten mit starken Magnetfeldern oder starken Funksignalen.
□ Installieren Sie das Gerät nicht an Orten, an denen statische Elektrizität auftreten kann. Wenn
die relative Luftfeuchtigkeit 45 % oder weniger beträgt, können sich Kunststoff- und
Isoliermaterialien durch Reibung usw. statisch aufladen.
□ Eine instabile Wechselstromversorgung kann zu Fehlfunktionen führen.
□ Entfernen Sie vor dem Gebrauch die Schutzfolie von der Waagschale.
□ Die Empfindlichkeit sollte bei der Erstinstallation der Waage oder nach einem längerenStandortwechsel eingestellt werden, um ein genaueres Wiegen zu ermöglichen. Weitere
Informationen finden Sie in der Bedienungsanleitung der FG-CWP/FG-ACWP-Serie auf
unserer Website https://www.aandd.jp.

2.2. Vorsichtsmaßnahmen für die Lagerung□ Nehmen Sie die Waage nicht auseinander.
□ Wenn die Waage verschmutzt ist, wischen Sie sie mit einem weichen Tuch ab, das leicht mit
einem milden Reinigungsmittel oder Reinigungsalkohol (70 % oder weniger) angefeuchtet ist.
Verwenden Sie keine organischen Lösungsmittel.
□ Schrubben Sie nicht mit einer Bürste oder ähnlichem.
□ Spritzen Sie sie nicht mit einem starken Wasserstrahl ab.
18

DE

2.3. Vorsichtsmaßnahmen zum Wiegen□ Legen Sie keine Lasten auf die Waagschale, die die Wägekapazität überschreiten.
□ Es wird empfohlen, die Empfindlichkeit regelmäßig zu justieren, um ein genaues Wiegen zu
gewährleisten.
□ Üben Sie keine Stöße auf die Waagschale aus und lassen Sie keine Gegenstände darauf fallen.
□ Verwenden Sie keine scharfen Gegenstände wie Bleistifte oder Stifte, um Tasten oder
Schalter zu drücken.
□ Drücken Sie vor jeder Wägung die ZERO -Taste, um die Fehlerwahrscheinlichkeit zu verringern.
□ Wiegen Sie nicht, wenn die Waage in Wasser getaucht ist.
□ Überprüfen Sie regelmäßig, ob die Wägewerte korrekt sind.
□ Um die Waage staub- und wasserdicht zu machen, sollte die Klappe auf der Rückseite derAnzeigeeinheit während der Verwendung geschlossen bleiben.

3. Mitgelieferte Artikel
Die folgenden Teile sind im Lieferumfang des Produkts enthalten.
< FG-CWP-Serie (USB-Stromversorgungsmodell)>● Hauptgerät
FG-30KCMWP
FG-60KCLWP
FG-150KCLWP● Schnellstartanleitung● USB-Kabel(Ca.3m)● Wechselstromadapter(Bilddiagramm)< FG-ACWP-Serie (AC-Modell)>● Hauptgerät
FG-30KACMWP
FG-60KACLWP
FG-150KACLWP● AC-Kabel(Ca. 3m)

19● Schnellstartanleitung

DE

4. Teilebezeichnungen< FG-CWP-Serie (USB-Stromversorgungsmodell)>Anzeigeeinheit (SUS304)
Typenschild
Seriennummer

USB-Kabel (Ca. 3 m)
USB-Stecker
Waagschale(SUS304)

Stange(SUS304)

Regler zur
Winkelanpassung

Ladezellenkabel
LibelleBlase

Basiseinheit(SUS304)

OK

Rückseite derAnzeigeeinheit

Nicht

Nivellierungsfüße< FG-ACWP-Serie (AC-Modell)>Typenschild
Seriennummer

Regler zur
Winkelanpassung

Anzeigeeinheit (SUS304)
AC-Kabel（Ca. 3m）
Stange（SUS304）

AC-Stecker

AC-kabel（Ca. 3m）

Weighing Pan（SUS304）

Ladezellen
kabel

Wägezellenpanel
LibelleBlasel

Basiseinheitt（SUS304）

OK

Nicht

Erde
Terminal
Erde

Rückseite derAnzeigeeinheit

Nivellierungsfüße

4.1. Anzeige und Symbole
Anzeigeeinheit

WiegezustandVergleichs-LED

Zustände fürjede Funktion

Wägeeinheiten

20

DE

Anzeige/Symbol
STABLE 〇NET
ZERO
Wägeeinheiten

Beschreibung
Dies leuchtet, wenn der Wägewert stabil ist, was anzeigt, dass die
Waage in einem ordnungsgemäßen Zustand zum Ablesen von
Wägewerten ist.
Dies leuchtet, wenn das Gewicht des Behälters (Tara) abgezogen wird.
Dies leuchtet, wenn sich die Waage am Nullpunkt befindet(Referenzpunkt für das Wiegen).
„kg“, „g“, „pcs“, „oz“ und „lb“ leuchten.
Hinweis: „g“ wird nur bei FG-30KCMWP/FG-30KACMWP verwendet.
Dies leuchtet, wenn die Anzeige gehalten wird.
Dies ist eine Funktion zum Erkennen eines Stoßes bzw. Aufpralls auf
den Massensensorbereich und zur Anzeige des Stoßpegels.
Leuchtet auf, wenn die optionale FG-27CWP (BluetoothKommunikationsschnittstelle) erfolgreich angeschlossen wurde.
Während die Vergleichsfunktion verwendet wird, wird der Wägewert
mit den voreingestellten Schwellenwerten verglichen und das Ergebnis
auf dem Bildschirm angezeigt.
Wechselstrom.
Erdungsanschluss.

□ Einzelheiten zur Funktionsweise und verschiedenen Funktionen finden Sie in derBedienungsanleitung der FG-CWP/FG-ACWP-Serie auf unserer Website
https://www.aandd.jp.
□ Bitte beachten Sie jedoch, dass Funktionen im Zusammenhang mit USB und mobilem Akku
nicht mit FG-ACWP-Serie (Wechselstrommodell) genutzt werden können.

4.2. Bedientasten
Bedientaste

Beschreibung
ON/OFF-Taste
Blendet die Anzeige ein oder aus.
Hinweis: Wenn der Netzstecker der Waage angeschlossen ist, werden die
Stromkreise im Inneren der Waage mit Strom versorgt.
SAMPLE-Taste

Durch die Auswahl von „pcs“ als Einheit wird der individuelle
Gewichtseinstellungsmodus aktiviert.
Durch Gedrückthalten (für etwa 2 Sekunden) wird der VergleichsSchwellenwerteinstellmodus aktiviert.
PRINT-Taste
Gibt die Wägewerte als Daten aus.
Erhöht beim Einstellen den Wert der blinkenden Ziffer.
MODE-Taste
Schaltet die Wägeeinheit um.
Verschiebt beim Einstellen die blinkende Ziffer nach rechts.
ZERO-Taste
Basierend auf dem Nullpunkt, der zum Zeitpunkt des Einschaltens derAnzeige gewogen wurde, schaltet sich der Nullpunktindikator durch Drücken
der ZERO-Taste ein, wenn der Gewichtswert innerhalb von ±2 % des
Wägebereichs stabil ist, sobald die Anzeige Null wird.

21

DE

TARE-Taste
Das Gewicht des Gegenstands auf der Waagschale wird als Gewicht eines
Behälters abgezogen (Tara-Gewicht).

5. Einrichtung
1.

Nehmen Sie die Basiseinheit und die Stange
aus dem Verpackungskarton und achten Sie
darauf, nicht am Kabel der Ladezellenkabel zu
ziehen.

Anzeigeeinheit

2.
Stange2. Entfernen Sie die Waagschale von der Basis.

3.

Zusammenbauen der Stange
Entfernen Sie vorübergehend die Schrauben
an der Unterseite der Wägebasis undbefestigen Sie die Stange und die Basis mit
den entfernten Schrauben, wobei Sie darauf
achten, dass das Ladezellenkabel nicht
eingeklemmt wird. Bündeln Sie dasüberschüssige Ladezellenkabel und führen Sie
es von der Displayseite her in die Stange ein.

Waagschale

1.
Basiseinheit

Ladezellenkabel

4. Platzieren Sie die Waagschale auf derBasiseinheit.

3.
Wenn Sie die Waage an einem Ort verwenden, an
dem leicht statische Elektrizität erzeugt wird oder
wo sie nass werden kann, befestigen Sie das
Erdungskabel mit der Erdungsklemme (M4Schraube).
Schutzerdungsklemme
Rückseite der Anzeigeeinheit

6.

Wartung

・Informationen zur Empfindlichkeitseinstellung finden Sie in der Bedienungsanleitung der FGCWP/FG-ACWP-Serie auf unserer Website https://www.aandd.jp.
・Überprüfen Sie regelmäßig die Genauigkeit des Wiegens. Justieren Sie die Waage, wenn sie aneinen anderen Ort gebracht wird oder sich die Umgebung verändert hat.

6.1.

Reparatur

Nehmen Sie die Waage nicht ohne einen autorisierten Servicetechniker auseinander oder bauen
Sie sie zusammen. Andernfalls besteht die Gefahr eines Stromschlags oder einer Beschädigung
der Waage usw. In diesem Fall ist eine Reparatur nicht durch die Garantie abgedeckt. Wenden Sie
sich an Ihren lokalen A&D-Händler, wenn Ihre Waage gewartet oder repariert werden muss.

22

DE

6.2.

Fehleranzeige

In dieser Situation
Nichts wird angezeigt.
Die Waage schaltet sichnicht ein.
Die Waage zeigt anfangs
nicht Null an.

Prüfen Sie diese Elemente
・Hat das Netzkabel die richtige Spannung?
・Ist das Netzkabel richtig angeschlossen?
・Prüfen Sie die Umgebung der Waagschale.
・Befindet sich etwas auf der Waagschale?
・Führen Sie die Nullpunkt-Empfindlichkeitsanpassung durch.

wird angezeigt.

・Wiegefehler, der „Überladen“ bedeutet.

wird angezeigt.

・Zeigt an, dass das Probengewicht zu gering ist, um das
Stückgewicht im Zählmodus einzustellen.

wird angezeigt.

・Empfindlichkeitsanpassungsfehler, der „Zu leicht“ bedeutet.

wird angezeigt.

・Wird angezeigt, wenn der Hauptstrom absinkt.

wird angezeigt
und der Vorgang wird nicht
fortgesetzt.
Feste Anzeige.

6.3.

・Der Wägewert ist aufgrund von Drift, Vibration oderanderen Faktoren instabil.
Eine Brise oder Vibration kann die Messung beeinträchtigen.
Prüfen Sie die Umgebung der Waagschale.
・Prüfen Sie den Anschluss des Ladezellenkabels.
・Haben Sie die „Haltefunktion“ verwendet?
Ändern Sie die Funktion.
・Schalten Sie die Waage aus und wieder ein.

Fehlercodetabelle

Wenn einer der folgenden Fehler angezeigt wird, versuchen Sie, die Anzeige aus- und wieder einzuschalten.

Anzeige

Inhaltwird angezeigt

Der Temparatursensor ist ausgefallen.

wird angezeigt

Der Speicher (Schaltkreis) ist ausgefallen.

wird angezeigt

Der interne Schaltkreis ist ausgefallen.

wird angezeigt

Der Massensensor ist ausgefallen.

7. Spezifikationen
ModelleWägekapazitätkg

Lesbarkeit
WägekapazitätgLesbarkeit
WägekapazitätlbLesbarkeit

FG-30KCMWP
FG-30KACMWP
30
0,010,005 *
0,002
0,001
30.000
10
5*
2
1
66
0,02
0,01 *
0,005
0,002
23

FG-60KCLWP
FG-60KACLWP
60
0,020,01 *
0,005
0,002

FG-150KCLWP
FG-150KACLWP
150
0,05
0,02 *
0,01
0,005

130
0,05
0,02 *
0,01
0,005

330
0,1
0,05 *
0,02
0,01

DE

ModelleWägekapazität

oz

lb-oz

Lesbarkeit
Wägekapazität
Lesbarkeit

Anzahl der Proben
Maximale Anzahl
Minimale Masse der Einheit [kg]Wiederholbarkeit(Standardabweichung) [kg]Linearität [kg]Stabilisierungszeit(gute Umgebung)
Temperaturdrift
Anzeige
Anzeigeaktualisierung
Schutz vor Staub und Wasser
BetriebsbedingungenInstallationsumgebung
Höhe

StromzufuhrÜberspannungskategorie **Verschmutzungsgrad **Größe der Waagschale [mm]Abmessungen [mm]Breite x Tiefe x Höhe

Gewicht (ungefähr)

FG-30KCMWP
FG-60KCLWP
FG-150KCLWP
FG-30KACMWP
FG-60KACLWP
FG-150KACLWP
1.050
2.100
5.200
0,5
1
2
0,2 *
0,5 *
1*
0,1
0,2
0,5
0,05
0,1
0,2
66
130
330
0,1
1
1
5 Stk. (kann zu 10 Stk., 20 Stk., 50 Stk. oder 100 Stk. geändert
werden)300.000 Stk.
0,00010,00020,0005
0,005

0,01

0,02

±0,01

±0,02
±0,05
20 kg: ca. 1,0 Sek. (Werkseinstellung)ca. 0,8 Sek. (Einstellung „“)±20 ppm/°C typ. (5 °C bis 35 °C )7-Segment-LCD mit Hintergrundlicht (Zeichenhöhe 30 mm)
Ca. 10 Mal/Sekunde
Entspricht der Schutzklasse IP67
-10 °C bis 40 °C, 85 % rel. Luftfeuchtigkeit oder weniger (nicht
kondensierend)
Nur Innen benutzen
Höhenlage bis 2.000 m<FG-CWP-Serie (USB-Stromversorgungsmodell)>Wechselstromadapter oder mobile Batterie oder USB-Port
Länge des USB-Kabels: Ca. 3 m(Das USB-Kabel wird auch für die Kommunikation verwendet.)< FG-ACWP-Serie (AC-Modell)>Wechselstromnetz (100 V bis 240 V, 50/60 Hz, 0,1 A max.)
AC-Kabellänge: Ca 3 m
II
2
300 x 380
370 x 500
300 x 515 x 784

370 x 635 x 784

11,1 kg
24,5 lb
30 kg
60 lb

12,8 kg
28,3 lb

Empfindlichkeitsanpass-ungs
60 kg
150 kg120 lb300 lbgewicht (Werkseinstellungen)*Werkseinstellungen** Gilt nur für FG-ACWP-Serie (Wechselstrommodell).
Spezifikationen für die Modelle der Klasse Ⅲ der FG-CWP/FG-ACWP-Serie
FG-30KCMWP
FG-60KCLWP
FG-150KCLWP
ModelleFG-30KACMWP
FG-60KACLWP
FG-150KACLWPGenauigkeitsklasseⅢ
Maximale Kapazität
30
60
150
kg
Lesbarkeit
0.01
0.02
0.05
Maximale Kapazität
66
130
330
lb
Lesbarkeit
0.02
0.05
0.1
Maximale Kapazität
1,050
2,100
5,200
oz
Lesbarkeit
0.5
1
2
0.2
0.4
1
Minimale Masse der Einheit [kg]
30
60
150
Maximale Tara [kg]□ Zugelassene Modellspezifikationen der Klasse Ⅲ. Einzelheiten finden Sie in derBedienungsanleitung der FG-CWP/FG-ACWP-Serie auf unserer Website
https://www.aandd.jp. Das Erscheinungsbild entspricht dem der FG-CWP/FG-ACWP-Serie.
24

ES

1. Introducción
En esta guía de inicio rápido se describen las funciones y operaciones básicas de la escala.
Las funciones y los procedimientos de uso de la báscula (serie FG-CWP/FG-ACWP) se explican
detalladamente en el manual de instrucciones siguiente.
[Manual de instrucciones de la serie FG-CWP/FG-ACWP]
Este es un manual de instrucciones para comprender y aprovechar al máximo la báscula deplataforma digital impermeable de la serie FG-CWP/FG-ACWP. Este manual se puede descargar
de nuestro sitio web https://www.aandd.jp.

Escanee este código QR para obtener este manual.

2. Precaución
2.1. Medidas de precaución para la instalación de la báscula
PELIGRO□ Conecte a tierra la báscula para que no haya peligro de descarga eléctrica al usuario.
□ Para evitar descargas eléctricas, no manipule el cable USB con las manos mojadas.
□ El cable USB no es resistente al agua. Utilice una toma USB ubicada en un lugar donde elenchufe no se moje. (Para la serie FG-CWP (modelo de alimentación USB))□ El enchufe de CA no es resistente al agua. Utilice un tomacorriente ubicado en un lugar
donde el enchufe no se moje. (Para la serie FG-ACWP (modelo de alimentación de CA))
□ No coloque la báscula en lugares donde haya gases inflamables o corrosivos.
□ No ejerza demasiada fuerza sobre los cables.
□ La báscula pesa bastante. Tenga cuidado al levantarla ya que podría caerse.
□ Use shielded load cell cables.

Para obtener el mejor rendimiento de su báscula, tenga en cuenta lassiguientes condiciones de instalación.
□ Las condiciones ideales de instalación son: temperatura y humedad estables, superficie
robusta y nivelada, lugar sin corrientes de aire ni vibraciones, ubicación interior y alejada dela luz solar directa, y fuente de alimentación estable.
□ No la coloque en un suelo blando o donde haya vibraciones.
□ No la coloque donde haya viento o grandes cambios de temperatura.
□ Evite zonas expuestas a la luz solar directa.
□ No la coloque en lugares con campos magnéticos fuertes o señales de radio potentes.
□ No la coloque en lugares donde pueda generarse electricidad estática. Cuando la humedad
relativa es del 45 % o menos, es más probable que el plástico y los materiales aislantes se
carguen con electricidad estática debido, por ejemplo, a la fricción.
□ Las fuentes de alimentación de CA inestables pueden causar averías.
□ Retire la película protectora del plato de pesaje antes del uso.
□ Para que el resultado del pesaje sea más preciso, el ajuste de sensibilidad debe realizarse al
instalar por primera vez la báscula o al trasladarla. Consulte el manual de instrucciones de laserie FG-CWP/FG-ACWP en nuestro sitio web https://www.aandd.jp.

2.2. Medidas de precaución para el almacenamiento□ No desmonte la báscula.
□ Si la báscula se ensucia, límpiela con un paño ligeramente humedecido con detergente
suave o con alcohol para frotar (70% o menos). No utilice disolventes orgánicos.
□ No frote con objetos como un cepillo.
□ No pulverice agua a presión.
25

ES

2.3. Medidas de precaución para el pesaje□ No coloque sobre el plato de pesaje cargas que superen la capacidad de pesaje.
□ Para garantizar la precisión de los resultados de pesaje se recomienda realizar el ajuste desensibilidad de forma periódica.
□ Evite que el plato de pesaje reciba golpes o que caigan cosas sobre él.
□ No utilice instrumentos afilados como un lápiz o un bolígrafo para pulsar las teclas o losinterruptores.
□ Pulse la tecla ZERO antes de cada pesaje para que haya menos probabilidad de error.
□ No utilice la báscula sumergida en agua cuando vaya a pesar.
□ Compruebe de forma periódica que los valores de pesaje son correctos.
□ El panel de la parte trasera de la unidad de visualización debe permanecer cerrado durante
el uso de la báscula para que sea resistente al polvo y al agua.

3. Elementos incluidos
Los elementos siguientes se proporcionan con el producto.
< Serie FG-CWP (modelo de alimentación USB)>● Unidad principal
FG-30KCMWP● Guía de inicio rápido
FG-60KCLWP
FG-150KCLWP●Cable USB(aprox. 3 m)

● Adaptador de CA（Esquema de la imagen)< Serie FG-ACWP (modelo de alimentación de CA)>
● Unidad principal
FG-30KACMWP● Guía de inicio rápido
FG-60KACLWP
FG-150KACLWP● Cable de CA(aprox. 3m)

26

ES

4. Nombres de las piezas< Serie FG-CWP (modelo de alimentación USB)>
Unidad de visualización(SUS304)

Etiqueta declasificación

Cable USB (aprox. 3 m)
Columna(SUS304)

Conector USB
Plato de pesaje(SUS304)

Cable de lacélula de carga

Unidad base
(SUS304)

Perilla de ajustedel ángulo
Nivel deburbujaParte trasera dela unidad devisualización

Bien Mal
Ruedas niveladoras< Serie FG-ACWP (modelo de alimentación de CA)>

Etiqueta declasificación

Knob for Angle
Adjustment

Unidad de visualización (SUS304)
Cable de AC（aprox. 3m）
Columna（SUS304）

enchufe de CA
Plato de pesaje（SUS304）

Unidad base
（SUS304）

Cable de CA（aprox.3m）

Cable dela célulade carga

Load Cell Panel

Ruedas niveladoras

Terminal
de tierra

Nivel deburbuja

TierraBien

Parte trasera dela unidad devisualizaciónMal

4.1. Pantalla y símbolos
Unidad de visualización

Estado delpesajeLED del comparador

Estado decada función

Unidades de pesaje

27

ES

Visualización/símbolo

Descripción

STABLE 〇Se enciende cuando el valor de pesaje es estable para indicar que labáscula ha alcanzado el estado adecuado para la lectura de losvalores de pesaje.
Se enciende cuando se resta el peso del recipiente (tara).

NET
ZERO
Unidades de pesaje

Se enciende cuando la báscula está en el punto cero (punto dereferencia para el pesaje).
Se enciende “kg”, “g”, “pcs”, “oz” y “lb”.
Nota: “g” se usa únicamente en la FG-30KCMWP/FG-30KACMWP.
Se enciende cuando se sujeta la pantalla.
Esta función detecta un impacto en la parte del sensor de masa ymuestra el nivel del impacto.
Se enciende cuando se establece correctamente la conexión con el
FG-27CWP opcional (interfaz de comunicación Bluetooth).
Cuando se usa la función del comparador, el valor de pesaje secompara con los valores umbral predefinidos y el indicador muestra el
resultado.
Corriente alterna.
Terminal de tierra (tierra).

□ Para obtener más información sobre el método de funcionamiento y las distintas funciones,
Consulte el manual de instrucciones de la serie FG-CWP/FG-ACWP en nuestro sitio web
https://www.aandd.jp.
□ However, please note that functions related to USB and mobile battery cannot be used with
Serie FG-ACWP (AC power model).

4.2. Teclas de funcionamiento
Tecla defuncionamiento

Descripción
Tecla ON/OFF
Muestra u oculta la visualización.
Nota: Cuando se conecta el enchufe de la báscula, se suministra energía alos circuitos eléctricos del interior de la báscula.
Tecla SAMPLE
Al seleccionar “pcs” como la unidad, se activa el modo de ajuste de peso
individual.
Mantener pulsado (durante unos 2 segundos) activa el modo ajuste delumbral del comparador.
Tecla PRINT
Envía los valores de pesaje en forma de datos.
Aumenta el valor del dígito que parpadea cuando se ajusta.
Tecla MODE
Cambia la unidad de pesaje.
Desplaza a la derecha el dígito que parpadea cuando se realizan ajustes..
Tecla ZERO
Al pulsar la tecla ZERO cuando el valor de pesaje es estable dentro de ±2 %de la capacidad de pesaje, el indicador de punto cero se enciende cuando
en la pantalla se muestra cero (según el punto cero pesado al encender lapantalla).

28

ES

Tecla TARE
Resta el peso del elemento situado en el plato de pesaje o peso del recipiente(tara).

5. Instalación
1.

Saque la unidad base y la columna de la cajade embalaje con cuidado de no tirar del cablede la célula de carga.

Unidad devisualización2.
2. Retire el plato de pesaje de la base.

3.

Columna
Plato depesajeMontaje de la columna
Retire de forma provisional los tornillos
fijados a la parte inferior de la base depesaje y, a continuación, fije la columna y labase con los tornillos retirados teniendo
cuidado de no pellizcar el cable de la célula
de carga. Ate el exceso de cable de la célula
de carga e introdúzcalo en la columna desdeel lado de la pantalla.

1.

Unidad base

4. Coloque el plato de pesaje en la unidad base.
Retire la película protectora del plato antes deluso.

Cable de lacélula de carga

3.

Cuando la utilice en un lugar en el que es
probable que se genere electricidad estática o en
uno húmedo, una el cable a tierra al borne
protector de puesta a tierra (tornillo).
Borne protector
de puesta a tierra
Parte trasera de launidad de visualización

6.

Mantenimiento

・Consulte el manual de instrucciones de la serie FG-CWP/FG-ACWP en nuestro sitio web
https://www.aandd.jp para ajustar la sensibilidad.
・Compruebe la precisión de los resultados de pesaje de forma periódica. Realice el ajuste de labáscula si se ha movido a otra ubicación o si ha cambiado el entorno.

6.1.

Reparación

No desmonte la báscula ni vuelva a montarla sin un ingeniero del servicio técnico autorizado. Si lo
hace, podría provocar una descarga eléctrica o daños a la báscula, etc. En ese caso, la garantía
no cubrirá la reparación. Póngase en contacto con su proveedor local de A&D para cualquier tarea
de mantenimiento o reparación de su báscula.

29

ES

6.2.

Visualización de errores

Si se da la situación siguiente
Compruebe los elementos siguientes
La pantalla está en blanco. ・¿La tensión de alimentación principal es la adecuada?
La báscula no se・¿El cable de alimentación principal está bien conectado?
enciende.
La báscula no muestra
cero al principio.

・Compruebe el entorno del plato de pesaje.
・¿Hay algo en el plato de pesaje?
・Realice el ajuste de sensibilidad del punto cero.

Aparece・Error de pesaje que quiere decir “sobrecarga”.

.

Aparece.

・Indica que el peso de muestra es demasiado bajo para configurar elpeso de la unidad en el modo de recuento.

Aparece.

・Error de ajuste de sensibilidad que quiere decir “demasiado bajo”.

Aparece.

Apareceno continúa.

yPantalla fija.

6.3.

・Se muestra cuando se produce una caída en la alimentación principal.
・El plato de pesaje está inestable debido a un movimiento, una
vibración u otros factores.
Una brisa o una vibración pueden estar afectando a la medición.
Compruebe el plato de pesaje.
・Compruebe la conexión del cable de la célula de carga.
・¿Ha utilizado la función “Hold”?
Consulte el [Manual de instrucciones de la serie FG-CWP].
・Cambiar la función.
・Apague la báscula y vuelva a encenderla.

Tabla de códigos de error

Cuando aparezca cualquiera de los errores siguientes, pruebe a apagar la pantalla y volver a encenderla.

Contenido

Visualización
Aparece.

Se ha producido un error en el sensor de temperatura.

Aparece.

Se ha producido un error en la memoria (el circuito).

Aparece.

Se ha producido un error en el sistema de circuitos interno.
Se ha producido un error en el sensor de masa.

Aparece

7. Especificaciones
Modelos
Capacidad de pesajekg

Legibilidad
Capacidad de pesajegLegibilidad
Capacidad de pesajelbLegibilidad

FG-30KCMWP
FG-30KACMWP
30
0,010,005 *
0,002
0,001
30.000
10
5*
2
1
66
0,02
0,01 *
0,005
0,002
30

FG-60KCLWP
FG-60KACLWP
60
0,020,01 *
0,005
0,002

FG-150KCLWP
FG-150KACLWP
150
0,05
0,02 *
0,01
0,005

130
0,05
0,02 *
0,01
0,005

330
0,1
0,05 *
0,02
0,01

ES

Modelos
Capacidad de pesaje

oz

lb-oz

Legibilidad
Capacidad de pesaje
Legibilidad

Número de muestras
Número máximo de recuento
Masa mínima de la unidad [kg]Repetibilidad(desviación estándar) [kg]Linealidad [kg]Tiempo de estabilización (buenentorno)Deriva de temperatura

FG-30KCMWP
FG-60KCLWP
FG-150KCLWP
FG-30KACMWP
FG-60KACLWP
FG-150KACLWP
1.050
2.100
5.200
0,5
1
2
0,2 *
0,5 *
1*
0,1
0,2
0,5
0,05
0,1
0,2
66
130
330
0,1
1
1
5 pcs (se puede cambiar a 10 pcs, 20 pcs, 50 pcs o 100 pcs)
300.000 pcs
0,00010,00020,0005
0,005

0,01

0,02

±0,01

±0,02
±0,05
20 kg: aprox. 1,0 s (ajuste de fábrica)
Aprox. 0,8 s ()±20 ppm/°C típ. (5 °C a 35 °C )LCD de 7 segmentos con retroiluminación
Pantalla(alto de los caracteres: 30 mm)
Aprox. 10 veces/segundo
Actualización de la pantalla
Cumple con IP67
Protección frente al polvo y el agua
-10 °C a 40 °C, 85 % HR o menos (sin condensación)
Condiciones de funcionamiento
Solamente para uso en interiores
Entorno de instalación
Altitud hasta 2.000 m
Altitud<Serie FG-CWP (USB power model)>Adaptador de CA, batería portátil o puerto USB
Longitud del cable USB: Aprox. 3 m(El cable USB también se utiliza para la comunicación).
Fuente de alimentación<Serie FG-ACWP (modelo de alimentación de CA)>Longitud del cable de CA (100 V a 240 V, 50/60 Hz, 0,1 A máx.
Longitud del cable de CA: aprox. 3 m
IICategoría de sobrevoltaje **2
Grado de contaminación **300 x 380
370 x 500
Tamaño del plato de pesaje [mm]Dimensiones [mm]300 x 515 x 784
370 x 635 x 784
Anchura x Profundidad x Altura
11,1 kg / 24,5 lb
12,8 kg / 28,3 lb
Peso (aproximado)
Peso de ajuste de sensibilidad
30 kg / 60 lb
60 kg / 120 lb
150 kg / 300 lb(ajuste predeterminado)* Ajuste predeterminado** Aplicable únicamente a Serie FG-ACWP (modelo de alimentación de CA)
Especificaciones de los modelos de la Serie FG-CWP/FG-ACWP de clase Ⅲ
FG-30KCMWP
FG-60KCLWP
FG-150KCLWP
Modelos
FG-30KACMWP
FG-60KACMWP
FG-150KACMWP
Clase de precisión
Ⅲ
Capacidad máxima
30
60
150
kg
Legibilidad
0.01
0.02
0.05
Capacidad máxima
66
130
330
lb
Legibilidad
0.02
0.05
0.1
Capacidad máxima
1,050
2,100
5,200
oz
Legibilidad
0.5
1
2
0.2
0.4
1
Masa mínima de la unidad [kg]
30
60
150
Tara máxima [kg]□ Especificaciones del modelo aprobado clase Ⅲ. Para obtener más información, consulte el
manual de instrucciones de la serie FG-CWP/FG-ACWP en nuestro sitio web
https://www.aandd.jp. La apariencia es la misma que la de la serie FG-CWP/FG-ACWP.
31

IT

1. Introduzione
Questa guida introduttiva descrive le funzioni di base e le operazioni della scala.
Le funzioni e le procedure operative della bilancia (Serie FG-CWP/FG-ACWP) sono illustrate nei
dettagli nel seguente manuale di istruzioni.
[Manuale di istruzioni della serie FG-CWP/FG-ACWP]
Questo è un manuale di istruzioni per comprendere e utilizzare appieno la bilancia digitale apiattaforma impermeabile serie FG-CWP/FG-ACWP. Questo manuale può essere scaricato dal
nostro sito web https://www.aandd.jp.
Eseguire la scansione di questo codice QR per questo manuale.

2. Attenzione
2.1. Precauzioni per l'installazione della bilancia
PERICOLO□ Tenere a terra la bilancia per evitare di esporre l'utente a scosse elettriche.
□ Per evitare scosse elettriche, non maneggiare il cavo USB con le mani bagnate.
□ Il cavo USB non è resistente all'acqua. Utilizzare una presa USB situata in un luogo in cui laspina non si bagni. (Per la serie FG-CWP (modello con alimentazione USB))□ La spina CA non è resistente all'acqua. Utilizzare una presa elettrica situata in un luogo in cui
la spina non si bagni. (Per la serie FG-ACWP (modello con alimentazione CA))□ Non applicare eccessiva forza sui cavi.
□ La bilancia è pesante. Prestare attenzione quando la si solleva, in quanto potrebbe cadere.
□ Utilizzare cavi schermati per celle di carico.

Tenere in considerazione le seguenti condizioni di installazione per ottenere il
miglior rendimento dalla bilancia.
□ Le condizioni ideali di installazione sono temperatura e umidità stabili, una superficie solida e
livellata, un luogo privo di correnti d'aria o vibrazioni, ambienti interni non direttamente
esposti alla luce solare e luoghi dotati di alimentazione stabile.
□ Non installare su un pavimento morbido o in presenza di vibrazioni.
□ Non installare in presenza di vento e grandi variazioni di temperatura.
□ Evitare i luoghi direttamente esposti alla luce solare.
□ Non installare in luoghi con campi magnetici notevoli o segnali radio forti.
□ Non installare in luoghi in cui potrebbe essere presente elettricità statica. In presenza diumidità relativa pari o inferiore al 45%, i materiali plastici e isolanti sono inclini a caricarsi dielettricità statica a causa della frizione, ecc.
□ Un'alimentazione CA instabile potrebbe causare malfunzionamenti.
□ Rimuovere la pellicola protettiva dal piatto per la pesatura prima dell'uso.
□ Al momento della prima installazione della bilancia o quando la si sposta lontano, è
opportuno eseguire una regolazione della sensibilità per una pesatura più precisa. Fareriferimento al manuale di istruzioni della serie FG-CWP/FG-ACWP sul nostro sito Web
https://www.aandd.jp.

2.2. Precauzioni per la conservazione□ Non smontare la bilancia.
□ Se la bilancia si sporca, pulirla con un panno morbido leggermente inumidito con un
detergente delicato o con alcol disinfettante (al 70% o meno). Non utilizzare solventi organici.
□ Non strofinare con una spazzola o oggetto simile.
□ Non sottoporre a forti getti d'acqua.
32

IT

2.3. Precauzioni per la pesatura□ Non collocare carichi che superino la capacità di pesatura sul piatto per la pesatura.
□ Si consiglia di eseguire regolazioni periodiche della sensibilità al fine di preservare la
precisione della pesatura.
□ Non scuotere il piatto per la pesatura ed evitare di farvi cadere oggetti.
□ Non utilizzare uno strumento appuntito, come una matita o una penna, per premere tasti ointerruttori.
□ Premere il tasto

ZERO

prima di ogni pesatura per ridurre le probabilità di errori.

□ Non effettuare la pesatura con la bilancia immersa in acqua.
□ Verificare periodicamente che i valori di pesatura siano corretti.
□ Il pannello sul retro dell'unità display deve essere tenuto chiuso durante l'utilizzo dellabilancia al fine di garantire la sua resistenza all'acqua e alla polvere.

3. Elementi inclusi
I seguenti elementi sono inclusi nella confezione del prodotto.
< Serie FG-CWP (modello con alimentazione USB)>● Unità principale● Guida introduttiva
FG-30KCMWP
FG-60KCLWP
FG-150KCLWP● Cavo USB(Circa 3m)● Adattatore AC（Illustrazione)< Serie FG-ACWP (modello con alimentazione CA)>
● Unità principale
FG-30KACMWP
FG-60KACLWP
FG-150KACLWP● Cavo CA(Circa. 3m)

33● Guida introduttiva

IT

4. Nomi dei componenti< Serie FG-CWP (modello con alimentazione USB)>
Unità display(SUS304)
Etichetta dellapotenza

Cavo USB (Circa 3 m)
Asta
(SUS304)

Spina USB
Piattoper la pesatura(SUS304)

Rotella per laregolazionedell'angolo

Cavo dellacella di carico
Livellaa bolla d’aria

Unità base
(SUS304)

OK

Retrodell'unità display

NO

Piedino di livellamento< Serie FG-ACWP (modello con alimentazione CA)>
Rotella per la regolazione
Unità display (SUS304)
dell'angolo

Etichetta dellapotenza

Cavo CA（Circa 3m）
Spina CA
Piattoper la pesatura（SUS304）

Unità base
（SUS304）

Asta（SUS304）
Cavo
della celladi carico

Cavo CA（Circa 3m）
Load Cell Panelterminale diterra

Livellaa bolla d’aria

Terra

OK

Retro
dell'unità display

NO

Piedino di livellamento

4.1. Display e simboli
Unità display

Condizionedella pesatura
Comparatore a LED

Condizioni perciascuna funzione

Unità di pesatura

34

IT

Display/simbolo

Descrizione

STABLE 〇Si accende quando il valore di pesatura è stabile; indica che la bilancia èin una condizione idonea alla lettura dei valori di pesatura.
Si accende quando si sottrae il peso del contenitore (tara).

NET
ZERO
Unità di pesatura

Si accende quando la bilancia è sul punto zero (punto di riferimento perla pesatura).
Si accende “kg”, “g”, “pcs”, “oz” e “lb”.
Nota: “g” viene utilizzato solo con il modello FG-30KCMWP /FG-30KACMWP.
Si accende quando si ferma il display.
Questa funzione rileva gli urti sulla sezione del sensore della massa evisualizza il livello di urto.
Si accende quando è stata stabilita con esito positivo la connessione
con la FG-27CWP (interfaccia di comunicazione Bluetooth) opzionale.
Durante l'utilizzo della funzione del comparatore, il valore di pesatura
viene confrontato utilizzando i valori di soglia preimpostati e l'indicatore
visualizza il risultato.
Corrente alternata.
Terminale di terra.

□ Per dettagli sul metodo di funzionamento e sulle varie funzioni, fare riferimento al manuale diistruzioni della serie FG-CWP/FG-ACWP sul nostro sito web https://www.aandd.jp.
□ However, please note that functions related to USB and mobile battery cannot be used with
Serie FG-D-ACWP (AC power model).

4.2. Tasti funzione
Tasto funzione

Descrizione
Tasto ON/OFF
Visualizza o nasconde il display.
Nota: quando la spina di alimentazione della bilancia è collegata, viene
fornita l’alimentazione ai circuiti elettrici all’interno della bilancia.
Tasto SAMPLE
Se si seleziona “pcs” come unità si attiva la modalità di impostazione delpeso singolo.
Tenendolo premuto (per circa 2 secondi) si attiva la modalità diimpostazione della soglia del comparatore.
Tasto PRINT
Emette il valore di pesatura come dato.
Aumenta il valore della cifra che lampeggia durante l’impostazione.
Tasto MODE
Commuta l'unità di pesatura.
Sposta verso destra la cifra che lampeggia quando si effettuano le impostazioni.
Tasto ZEROIn base al punto zero pesato al momento dell'accensione del display, se si
preme il tasto ZERO quando il valore del peso è stabile entro ±2% dellacapacità di pesatura, l'indicatore del punto zero si accende non appena il
display diventa zero.

35

IT

Tasto TARE
Il peso dell'articolo presente sul piatto per la pesatura viene sottratto comepeso del contenitore (peso della tara).

5. Configurazione
1.

Estrarre la base e l'asta dalla confezione
prestando attenzione a non tirare il cavo dellacella di carico.

Unità display

2.
2. Rimuovere il piatto per la pesatura dalla base.

3.

Montaggio dell'asta
Rimuovere temporaneamente le viti fissate
alla parte inferiore della base di pesatura, quindi
fissare l’asta e la base con le viti rimosse,
facendo attenzione che il cavo della cella dicarico non venga schiacciato.
Legare e inserire il cavo della cella di carico
eccedente nell'asta dal lato del display.

Asta

Piatto perla pesatura

1.

Base

Cavo dellacella di carico

4. Posizionare il piatto per la pesatura sulla base.

3.
Quando si utilizza in un luogo in cui potrebbe
generarsi elettricità statica o in un luogo umido, unire
il cavo di messa a terra al morsetto di terra protettivo(vite).
Morsetto diterra protettivo
Retro dell'unità display

6.

Manutenzione

・Fare riferimento al manuale di istruzioni della serie FG-CWP/FG-ACWP sul nostro sito Web
https://www.aandd.jp per la regolazione della sensibilità.
・Verificare periodicamente la precisione di pesatura. Regolare la bilancia a seguito di uno
spostamento in un luogo diverso o se l'ambiente è cambiato.

6.1.

Riparazione

Non smontare/montare la bilancia se non in presenza di un tecnico di assistenza autorizzato. In
caso contrario, si potrebbero causare scosse elettriche o danni alla bilancia, ecc. In tali evenienze,la riparazione non è coperta dalla garanzia. Se si ha bisogno di assistenza o di una riparazione perla propria bilancia, contattare il rivenditore locale di A&D.

36

IT

6.2.

Visualizzazione degli errori

In questa situazione
Verificare questi aspetti
・La tensione dell'alimentazione principale è quella corretta?
Nessuna visualizzazione.
La bilancia non si accende. ・Il cavo dell'alimentazione principale è collegato adeguatamente?
・Controllare intorno al piatto per la pesatura.
La bilancia non visualizza・C'è qualcosa sul piatto per la pesatura?
lo zero all'inizio.
・Eseguire la regolazione della sensibilità del punto zero.
・Errore di pesatura che significa “Sovraccaricato”.
Si visualizza
・Indica che il peso del campione è troppo leggero per impostare
Si visualizza
il peso dell'unità nella modalità di conteggio.
・Errore di regolazione della sensibilità che significa “Troppo leggero”.
Si visualizza
・Appare quando scende l'alimentazione principale.

Si visualizza

Si visualizzael’operazione non procede.

Display fisso.

6.3.

・Il valore pesato è instabile a causa di spostamenti, vibrazioni
o altri fattori.
Una brezza o delle vibrazioni potrebbero stare influenzando
la misurazione.
Controllare l’area circostante il piatto di pesatura.
・Controllare il collegamento del cavo della cella di carico.
・Si è usata la funzione “Hold”?
Cambia la funzione
・Spegnere la bilancia e riaccenderla.

Tabella dei codici di errore

Se si visualizza uno dei seguenti errori, provare a spegnere e riaccendere il display.

Display

Contenuto

Si visualizza

Guasto del sensore di temperatura.

Si visualizza

Guasto della memoria (circuito).

Si visualizza

Guasto del circuito interno.

Si visualizza

Guasto del sensore della massa.

7. Specifiche tecniche
Modelli
Capacità di pesaturakg

Leggibilità
Capacità di pesaturag

Leggibilità
Capacità di pesaturalb

Leggibilità

FG-30KCMWP
FG-30KACMWP
30
0,010,005 *
0,002
0,001
30.000
10
5*
2
1
66
0,02
0,01 *
0,005
0,002
37

FG-60KCLWP
FG-60KACLWP
60
0,020,01 *
0,005
0,002

FG-150KCLWP
FG-150KACLWP
150
0,05
0,02 *
0,01
0,005

130
0,05
0,02 *
0,01
0,005

330
0,1
0,05 *
0,02
0,01

IT

Modelli
Capacità di pesatura

oz

lb-oz

Leggibilità
Capacità di pesatura
Leggibilità

Numero di campioni
Numero massimo di conteggi
Massa unitaria minima [kg]Ripetibilità(Deviazione standard) [kg]Linearità [kg]Tempo di stabilizzazione(buon ambiente)
Deriva termica
Display
Aggiornamento del display
Protezione da polveri e acqua
Condizioni di funzionamento
Ambiente di installazione
Altitudine

Alimentazione

Overvoltage category **Pollution degree **Dimensioni del piatto per lapesatura [mm]Dimensioni [mm]Larghezza x Profondità x Altezza
Peso (circa)

FG-60KCLWP
FG-150KCLWP
FG-30KCMWP
FG-30KACMWP
FG-60KACLWP
FG-150KACLWP
1.050
2.100
5.200
0,5
1
2
0,2 *
0,5 *
1*
0,1
0,2
0,5
0,05
0,1
0,2
66
130
330
0,1
1
1
5 pezzi (può essere impostato su 10 pezzi, 20 pezzi, 50 pezzi o 100 pezzi)
300.000 pezzi
0,00010,00020,0005
0,005

0,01

0,02

±0,01
±0,02
±0,05
20 kg: circa 1,0 sec (impostazione di fabbrica)
Circa 0,8 sec (impostazione)±20 ppm/°C tip. (da 5 °C a 35 °C)
LCD a 7 segmenti con retroilluminazione (altezza del carattere 30 mm)
Circa 10 volte al secondo
Conforme IP67
Da -10 °C a 40 °C, max. 85% di umidità relativa (senza condensa)
Solo per uso interno
Altitudine fino a 2.000 m<Serie FG-CWP (modello con alimentazione USB)>
Adattatore CA o batteria mobile o porta USB
Lunghezza del cavo USB: circa 3 m(Il cavo USB viene utilizzato anche per la comunicazione.)<Serie FG-ACWP (modello con alimentazione CA)>
Rete CA (Da 100 V a 240 V, 50/60 Hz, 0,1 A massimo)
Lunghezza del cavo CA: circa. 3 m
II
2
300 x 380

370 x 500

300 x 515 x 784

370 x 635 x 784

11,1 kg
24,5 lb

12,8 kg
28,3 lb

Peso di regolazione della
30 kg
60 kg
150 kgsensibilità (impostazione di
60 lb
120 lb300 lbfabbrica)* Impostazione di fabbrica** Applicabile solo a Serie FG-ACWP (modello con alimentazione CA).
Specifiche tecniche del modello Serie FG-CWP/FG-ACWP Classe Ⅲ
FG-30KCMWP
FG-60KCLWP
FG-150KCLWP
Modelli
FG-30KACMWP
FG-60KACLWP
FG-150KACLWPⅢClasse di precisione
Capacità massima
30
60
150
kg
Leggibilità
0.01
0.02
0.05
Capacità massima
66
130
330
lb
Leggibilità
0.02
0.05
0.1
Capacità massima
1,050
2,100
5,200
oz
Leggibilità
0.5
1
2
0.2
0.4
1
Massa unitaria minima [kg]Tara massima [kg]
30
60
150
□ Specifiche del modello approvato di classe Ⅲ. Per i dettagli, fare riferimento al manuale diistruzioni della serie FG-CWP/FG-ACWP sul nostro sito Web https://www.aandd.jp. L'aspetto
è lo stesso della serie FG-CWP/FG-ACWP.
38

CT

1. 簡介
本快速入門指南描述秤的基本功能和操作。
秤的詳細功能及操作程序（FG-CWP/FG-ACWP 系列）將在下方使用手冊中說明。
[FG-CWP/FG-ACWP 系列使用手冊]
這是一本了解並充分利用防水數位平台秤 FG-CWP/FG-ACWP 系列的使用說明書。 本手冊可以從我們的網站 https://www.aandd.jp 下載。
掃描此二維碼即可取得本手冊。

2. 警告
2.1. 秤的安裝注意事項危險□ 請讓秤接地，以確保使用者不會觸電。
□ 為避免觸電，請勿濕手操作 USB 纜線。
□ USB 連接線不防水。 使用位於插頭不會受潮的地方的 USB 插座。（適用於 FG-CWP 系列（USB 電源型號））□ AC 插頭不防水。 使用位於插頭不會受潮的地方的電源插座。 （適用於 FG-ACWP 系列（交流電源型號））□ 請勿用力拉扯纜線。
□ 秤很重。抬起時可能導致秤翻倒，因此請小心。
□ 使用屏蔽稱重感測器電纜。

為讓秤達到最佳效能，選擇安裝地點時請考量下列安裝條件。
□ 理想的安裝條件為溫定的溫度及濕度、堅固且平坦的表面、不受風吹或震動的地點、遠離陽
光直射的室內，以及穩定的電源。
□ 請勿安裝在柔軟的地面或會震動處。
□ 請勿安裝在受強風吹拂或溫度波動大的地點。
□ 避免陽光直射的地點。
□ 請勿安裝在有強大磁場或強大無線電訊號的地點。
□ 請勿安裝在可能產生靜電的地點。濕度為 45% R.H. 或以下時，塑膠及絕緣材料容易因摩
擦等而帶靜電。
□ 使用不穩定的 AC 電源可能導致故障。
□ 使用前，請撕掉秤盤的保護膜。
□ 首次安裝秤或遠距離移動秤時，應執行敏感度調整，以讓稱重更加精準。請參閱我們網站
https://www.aandd.jp. 上的 FG-CWP/FG-ACWP 系列說明手冊。

2.2. 存放注意事項□ 請勿拆解秤。
□ 如果秤變髒，請使用蘸有溫和清潔劑或外用酒精（70% 或更少）的軟布輕輕擦拭。請勿使
用有機溶劑。
□ 請勿使用刷子或類似工具刷洗。
□ 請勿使用強力水柱噴水。

39

CT

2.3. 稱重注意事項□ 請勿將超出稱重上限的荷重置於秤盤上。
□ 建議定期執行敏感度調整，以維持稱重精準度。
□ 請勿撞擊或讓任何物品掉落在秤盤上。
□ 請勿用鉛筆或筆等尖銳物品來按按鍵或開關。
□ 每次稱重前按下 ZERO 按鍵，以減少發生錯誤的機會。
□ 請勿在水中用秤稱重。
□ 定期確認稱重值是否正確。
□ 使用秤時，顯示幕背面的面板應保持關閉，以便防塵與防水。

3. 內容物產品隨附下列項目。
< FG-CWP 系列（USB 供電型號）>● 主機
FG-30KCMWP● 快速入門指南
FG-60KCLWP
FG-150KCLWP● USB 纜線
(約。3m)● AC 變壓器

(示意圖）< FG-ACWP 系列（交流電源型）>● 主機
FG-30KACMWP● 快速入門指南
FG-60KACLWP
FG-150KACLWP● AC 纜線(約。3m)

40

CT

4. 零件名稱< FG-CWP 系列（USB 供電型號）>顯示幕(SUS304)
分級標籤序號

USB 纜線（約 3m）
立桿(SUS304)

USB 插頭秤盤(SUS304)

荷重元纜線

基座(SUS304)

氣泡式
水平儀

OK

角度調整旋鈕
背面
（顯示幕）

NG

水平調整腳座< FG-ACWP 系列（AC 電源型）>
顯示幕 (SUS304)

分級標籤序號

角度調整旋鈕

AC 纜線（約 3m）
立桿（SUS304）

AC 插頭

AC 纜線
（約. 3m）

秤盤（SUS304）

荷重元纜線

基座（SUS304）

水平調整腳座荷重元控制板
接地端子

氣泡式
水平儀

接地

OK

背面
（顯示幕）

NG

4.1. 顯示和符號
顯示幕

稱重
條件
比較器 LED

各項功能
條件

稱重單位

41

CT

顯示／符號
STABLE 〇描述稱重值穩定時亮起，表示秤處於可讀取稱重值的正常狀態。

NET

減去容器重量（皮重）時亮起。

ZERO秤為零點（稱重參考點）時亮起。

稱重單位

「kg」
、「g」、「pcs」、「oz」及「lb」亮起。
注意：「g」僅用於 FG-30KCMWP/FG-30KACMWP。

顯示內容固定時亮起。
此功能用於偵測稱重感應器部分受到的撞擊，並顯示撞擊力等級。
與 FG-27CWP (選購件) 成功連線時會亮起。
（藍牙通訊介面）。
使用比較器功能時，稱重值會與預設閾值進行比較，並由指示燈顯示結果。

交流電。
接地端子。

□ 有關操作方法和各種功能的詳細信息，請參閱我們網站 https://www.aandd.jp.上的 FGCWP/FG-ACWP 系列使用說明書。
□ 但請注意，FG-ACWP 系列（AC 電源型號）無法使用與 USB 和行動電池相關的功能。

4.2. 操作按鍵
操作按鍵

描述
ON/OFF 按鍵顯示或隱藏顯示內容。
注意：接上秤的電源插頭，會為秤內部的電路供電。
SAMPLE 按鍵選擇「pcs」作為單位後，啟動個別重量設定模式。
按住按鍵約 2 秒後，啟動比較器閾值設定模式。
PRINT 按鍵
輸出稱重值資料。
設定時，可增加閃爍數字的值。
MODE 按鍵切換稱重單位。
設定時，可切換到閃爍數字的右邊。
ZERO 按鍵在稱重值穩定維持在稱重上限的 ±2% 內時按下 ZERO 按鍵後，零點指示燈
即會根據顯示幕開啟時稱得的零點，在顯示變為零時立即亮起。

42

CT

TARE 按鍵將秤盤上物品的重量視為容器的重量減去（皮重）
。

5. 設定
1.

將基座和立桿從包裝盒取出，同時小心勿拉扯到
荷重元纜線。

顯示幕

立桿

2.

2. 將秤盤從基座拆下。

秤盤

3.

組裝立桿
暫時取下安裝在秤盤基座底部的螺絲，然後用取下的螺絲固定立桿和基座，注意不要夾住荷重元
纜線。將多餘的荷重元纜線綁好，從顯示幕側放
入立桿中。

1.

基座
荷重元纜線

4. 將秤盤置於基座上。

3.
如果在容易產生靜電或可能受潮的地方使用秤，
請用接地端子（M4 螺絲）固定接地線

保護接地端子
顯示幕背面

6.

維護

・有關靈敏度調整，請參閱我們網站 https://www.aandd.jp 上的 FG-CWP/FG-ACWP 系列使用說明
書。
・定期檢查稱重精準度。若將秤移至其他地點或環境有所改變，請進行調整。

6.1.

維修

請勿未經授權保養技師指示即拆解／組裝秤。否則可能導致觸電或秤受損等。此時，維修將不包含在保固範圍內。若您的秤需要保養或維修，請聯絡當地 A&D 經銷商。

43

CT

6.2.

顯示錯誤

在此情況下

確認項目

未顯示任何內容。
秤未開啟。

・主電源電壓是否正確？
・主電源線是否確實連接？

秤開始時未顯示零。

・檢查秤盤。
・秤盤上是否有東西？
・執行零點敏感度調整。

顯示。

・稱重錯誤，表示「過重」。

顯示。

・表示樣品重量太輕，無法在計數模式設定單位重量。

顯示。

・敏感度調整錯誤，表示「太輕」。

顯示。

・主電源下降時顯示。

顯示，且不會繼續運作。

・由於漂移、振動或其他因素，導致稱重值不穩定。
微風或振動可能會影響測量結果。
請檢查秤盤周圍。
・檢查稱重感應器纜線的連接。

顯示內容固定。

・是否使用「固定功能」？ 變更・關閉再重新開啟秤。

6.3.

功能。

錯誤代碼表

顯示下列任一錯誤時，請嘗試將顯示幕關閉再重新開啟。

顯示

內容

顯示

溫度感應器故障。

顯示

記憶體（電路）故障。

顯示

內部電路故障。

顯示

稱重感應器故障。

7. 規格
型號
稱重上限

kg

可讀數
稱重上限

g

可讀數
稱重上限

lb

可讀數

FG-30KCMWP
FG-30KACMWP
30
0.01
0.005 *
0.002
0.001
30,000
10
5*
2
1
66
0.02
0.01 *
0.005
0.002
44

FG-60KCLWP
FG-60KACLWP
60
0.02
0.01 *
0.005
0.002

FG-150KCLWP
FG-150KACLWP
150
0.05
0.02 *
0.01
0.005

130
0.05
0.02 *
0.01
0.005

330
0.1
0.05 *
0.02
0.01

CT

型號
稱重上限

oz

lb-oz

可讀數
稱重上限
可讀數

樣品數
最大計數
最小單位質量 [kg]可重複性（標準偏差）[kg]
線性 [kg]
穩定時間
（環境佳）
溫度漂移
顯示
顯示內容更新
防塵防水
操作條件
安裝環境
高度

電源

FG-30KCMWP
FG-60KCLWP
FG-150KCLWP
FG-30KACMWP
FG-60KACLWP
FG-150KACLWP
1,050
2,100
5,200
0.5
1
2
0.2 *
0.5 *
1*
0.1
0.2
0.5
0.05
0.1
0.2
66
130
330
0.1
1
1
5 pcs（可變更至 10 pcs、20 pcs、50 pcs 或 100 pcs）
300,000 pcs
0.00010.00020.0005
0.005
0.01
0.02
±0.01
±0.02
±0.05
20 公斤：約 1.0 秒（出廠設定）
約 0.8 秒（設定）
±20 ppm/°C typ.（5 °C 至 35 °C）
7 段 LCD 帶背光（字元高度 30 mm）
約 10 次／秒
符合 IP67-10 °C 至 40 °C，85% R.H. 或以下（無凝結）
僅限室內使用海拔高度可達 2,000m< FG-CWP 系列（USB 供電型號）>AC 變壓器或行動電池或 USB 連接埠
USB 纜線長度：大約 3 m(USB 纜線也用於通訊。）< FG-ACWP 系列（交流電源型號）>AC 電源 (100V 至 240V，50/60Hz，最大 0.1A)
AC 纜線長度: 約 3 m
II
2
300 x 380
370 x 500
300 x 515 x 784
370 x 635 x 784
11.1 kg / 24.5 lb
12.8 kg / 28.3 lb
30 kg / 60 lb
60 kg / 120 lb
150 kg / 300 lb

過電壓類別**
污染程度**
秤盤尺寸 [mm]
尺寸 [mm] 寬 x 深 x 高
重量（約略值）敏感度調整砝碼（出廠設定）*
出廠設定** 僅適用於 FG-ACWP 系列（AC 電源型號）FG-CWP/FG-ACWP 系列 Class Ⅲ 型號規格FG-30KCMWP
型號
FG-30KACMWP
精度等級
最大容量
30
kg
可讀數
0.01
最大容量
66
lb
可讀數
0.02
最大容量
1,050
oz
可讀數
0.5
最小單位質量 [kg]0.2
最大皮重 [kg]30

FG-60KCLWP
FG-60KACLWP
Ⅲ
60
0.02
130
0.05
2,100
1
0.4
60

FG-150KCLWP
FG-150KACLWP
150
0.05
330
0.1
5,200
2
1
150

□ Ⅲ類認可型號規格。 詳情請參閱本公司網站 https://www.aandd.jp 上的 FG-CWP/FG-ACWP
系列使用說明書。 外觀與 FG-CWP/FG-ACWP 系列相同。
45

Ｍemo

46

THIS PAGE INTENTIONALLY BLANK LEFT.

A&D Company, Limited

3-23-14 Higashi-Ikebukuro, Toshima-ku, Tokyo 170-0013, JAPAN
Telephone: [81] (3) 5391-6132
Fax: [81] (3) 5391-1566

A&D ENGINEERING, INC.

47747 Warm Springs Blvd, Fremont, California 94539, U.S.A.
Tel: [1] (800) 726-3364
Weighing Support:[1] (888) 726-5931

A&D INSTRUMENTS LIMITED

Inspection Support:[1] (855) 332-8815

Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire
Telephone: [44] (1235) 550420
Fax: [44] (1235) 550485

OX14 1DY

United Kingdom

A&D AUSTRALASIA PTY LTD

32 Dew Street, Thebarton, South Australia 5031, AUSTRALIA
Telephone: [61] (8) 8301-8100
Fax: [61] (8) 8352-7409

A&D KOREA Limited

한국에이.엔.디(주)OOO A&D RUS

OOO "ЭЙ энд ДИ РУС"서울특별시 영등포구 국제금융로6길33 (여의도동) 맨하탄빌딩 817 우편 번호 07331( 817, Manhattan Bldg., 33. Gukjegeumyung-ro 6-gil, Yeongdeungpo-gu, Seoul, 07331 Korea )전화: [82] (2) 780-4101
팩스: [82] (2) 782-4264

Почтовый адрес:121357, Российская Федерация, г.Москва, ул. Верейская, дом 17Юридический адрес: 117545, Российская Федерация, г. Москва, ул. Дорожная, д.3, корп.6, комн. 8б
( 121357, Russian Federation, Moscow, Vereyskaya Street 17 )тел.: [7] (495) 937-33-44
факс: [7] (495) 937-55-66

A&D Instruments India Private Limited

D-48( D-48, Udyog Vihar, Phase- , Gurgaon - 122016, Haryana, India ): [91] (124) 4715555: [91] (124) 4715599

A&D SCIENTECH TAIWAN LIMITED.

A&D台灣分公司 艾安得股份有限公司

台湾台北市中山區南京東路 2 段 206 號 11 樓之 2
( 11F-2, No.206, Sec.2, Nanjing E.Rd., Zhongshan Dist., Taipei City 10489, Taiwan, R.O.C. )Tel : [886](02) 2322-4722
Fax : [886](02) 2392-1794

A&D INSTRUMENTS (THAILAND) LIMITEDบริษัท เอ แอนด์ ดี อินสทรูเม้นท์ (ไทยแลนด์) จำกัด
168/16 หมู่ที่ 1 ตำบลรังสิต อำเภอธัญบุรี จังหวัดปทุมธานี 12110 ประเทศไทย

( 168/16 Moo 1, Rangsit, Thanyaburi, Pathumthani 12110 Thailand )Tel : [66] 20038911