-N - E-W-

Platform Scales

FGSeries

dnoce
S1ts
Ju
FG -150KAL

FG -30KAM

FG -60KBM

ISO9000 CERTIFIED

http://www.aandd.jp

FG-150KAL
FG-60KAL
FG-150KAM
FG-60KAM
FG-30KAM
FG-150KBM
FG-60KBM
FG-30KBM

Fast Weighing Res
for Efficient Weigh

The New FG series accomplishes fast weighing response time andexcellent feature of a three weighing resolution function enablesfrom Normal resolution of 1/3000, High resolution of 1/6000 (orany weighing demand.

dnoce
SJust 1E
STABL
NE TZEROicators
tor indarapomCFG -30KAM
FG -60KAM
FG -150KAM
FG -60KAL
FG -150KAL

FG -30KBM
FG -60KBM
FG -150KBM

OFFsponsehing Applications!

d accurate result to suit any weighing applications. Furthermore its
the user to select the weighing resolution in the Function setting
1/7500) and Higher resolution of 1/12000 ( or 1/15000), to meet

Three weighing resolutions
Selectable from Normal resolution (1/3000), High resolution (1/6000 for 30kg and 60kg weighing capacity) and Higherresolution (1/12000 for 60kg and 1/15000 for 30kg and 150kg weighing capacity) by function setting

Counting function
Useful in piece counting applications with sample size of 5, 10, 20, 50 and 100 pcs for unit weight registration

Comparator function
HI / OK / LO comparator indication and optional comparator relay output ( FG-24) for checkweighing applications

Calibration function
Easy and simple to calibrate the scale with external calibration mass for accurate weighing

Auto-power OFF function

OFFSaves power consumption by automatically switching itself off when the scale displays zero for approx. five minutes

Selectable two sizes of stainless steel weighing pan
Stainless steel weighing pan is easy to clean and weighing pan of size of M size (300 x 380mm) or L size (390 x530mm) can be selected

Two types of power supply
AC Adapter for permanent installation applications or dry batteries of 4 x C size (R14P/ LR14) for applications inlocations with no AC power supply

Low power supply indication function
Easy to know when power supply is low with the display of “ Lb1” for AC power supply and “ Lb0” for dry batteries

Filtering function
Filtering levels of Slow, Normal and Fast are selectable to suit weighing environment conditions

Leveling foot and Leveling bubble
Easy and simple to adjust the level of the scale for accurate weighing

Display column and Tilt display pod
The column and tilting angle of the display unit is adjustable for better reading of display

Status indicator function
The display has status indicators of Stable, Net and Zero for easy operation

Power– ON Zero function
Takes Zero as its center when power is ON for accurate weighing

Gravity acceleration correction function
Adjustable gravity acceleration for accurate weighing

Optional RS-232C and Comparator Output
Easy to connect peripherals such as printer, comparator light and computer for weighing system applications

FG Series Specifications
FG-30KAM
FG-30KBM

FG-60KAM
FG-60KBM

FG-150KAM
FG-150KBM

FG-60KAL FG-150KAL

Weight capacity (kg)306015060150

Min. display (kg)

0.01

0.02

0.05

0.02

0.05

0.005*

0.01*

0.02*

0.01*

0.02*

0.002

0.005

0.01

0.005

0.01

Weight capacity (lb)**

60

150300150300Min. display (lb)**

0.02

0.05

0.1

0.05

0.1

0.01*

0.02*

0.05*

0.02*

0.05*

Weight capacity (oz)**
Min. display (oz)**

0.005

0.01

0.02

0.01

0.0296024004800

2400

4800

0.5

12

12

0.2*

0.5*

1*

0.5*

1*

0.2

0.5

0.2

0.5
120,000 pcs

0.1

5 (can be changed to 10, 20, 50 or 100 ) pieces

No. of samples

120,000 pcs

96,000 pcs

120,000 pcs

96,000 pcs

Min. unit weight

0.25g

0.625g1.25g0.625g

1.25g

Repeatability (Std. deviation)

0.005kg

0.01kg0.02kg0.01kg0.02kgLinearity error±0.01kg±0.02kg±0.05kg±0.02kg±0.05kg

Max. counts±20ppm / °C (10°C~30°C / 50°F~86°F)

Sensitivity drift

7 segment LCD display (Character height 26mm)

Display
Display update

10 times per second

Operating temp

—10°C~40°C / 14°F~104°F, less than 85% RH (no condensation )AC adapter or C size (R14P / LR14) x 4 batteries

Power supply

Approximately 150 hours with alkaline dry cell battery

Battery operating

300 x 380mm / 11.8 x 15.0inches

390 x 530mm / 15.4 x 20.9inches

FG-KAM: 300(W) x 624(D) x 781(H)mm / FG-KBM: 380(W) x 464(D) x 118(H)mm

390(W) x 771(D) x 781(H)mm

Weighing pan size
Dimension
Calibration weight (Factory setting)

16.4kg

FG-KAM: 11.2kg / FG-KBM: 9.7kg

Weight (approximately)

30kg / 60lb

60kg / 150lb

150kg / 300lb

60kg / 150lb

150kg / 300lb

setting* Factory** U.S.A. version onlyOptions
FG-23
FG-24

RS-232C serial interface
RS-232C serial interface and comparator relay output

Accessories
AD-8121B
AD-8118B

Dot Matrix Compact Printer
Universal PrinterAD-8121B...Clearly a Better Value
3-23-14 Higashi-lkebukuro, Toshima-ku,Tokyo 170-0013 JAPAN
Telephone:[81](3) 5391-6132 Fax:[81](3) 5391-6148
http://www. aandd.jp

Unit 24/26 Blacklands Way Abingdon Business Park,Abingdon, Oxon OX14 1DY United Kingdom
Telephone:[44](1235) 550420 Fax:[44](1235) 550485< German Sales Office>1555 McCandless Drive, Milpitas, CA. 95035 U.S.A.
Telephone:[1](408) 263-5333 Fax:[1](408) 263-0119

Große Straße 13 b 22926 Ahrensburg GERMANY
Telephone:[49](0) 4102 459230 Fax:[49](0) 4102 459231

32 Dew Street, Thebarton, South Australia 5031 AUSTRALIA
Telephone:[61](8) 8301-8100 Fax:[61](8) 8352-7409

Manhattan Bldg. 8F, 36-2 Yoido-dong, Youngdeungpo-gu, Seoul, KOREA
Telephone:[82](2) 780-4101 Fax:[82](2) 782-4280* FG-ADCC-02-PR2-05605