INSTRUCTION MANUALDIGITAL CLAMP METERKEW SNAP SERIESKEW SNAP 2 0 0 2 ＰＡ 2 0 0 2 R
KYORITSU ELECTRICAL INSTRUMENTS
WORKS，LTD.

Contents

ENGLISH

１．Safety Warnings ……………………………………………………… 1
２．Features ……………………………………………………………… 4
３．Specifications
３−１ KEW SNAP 2002PA …………………………………………… 5
３−２ KEW SNAP 2002R …………………………………………… 7
４．Instrument Layout …………………………………………………… 10
５．Preparation for Measurement
５−１ Battery Voltage Check ……………………………………… 11
５−２ Checking Switch Setting and Operation …………………… 11
６．Measurement
６−１ Current Measurement………………………………………… 12
６−２ Voltage Measurement ……………………………………… 14
６−３ Resistance Measurement …………………………………… 15
７．Notes on Functions
７−１ Data Hold ……………………………………………………… 17
７−２ Sleep Function………………………………………………… 17
７−３ Output for Recorder ………………………………………… 18
７−４ Mode Switching Function …………………………………… 19
８．Battery Replacement ………………………………………………… 20
９．Disposing the Product ………………………………………………… 21

１．Safety Warnings○This instrument has been designed and tested according to IEC
Publication 61010；Safety Requirements for Electronic Measuring
Apparatus. This instruction manual contains warnings and safety
rules which must be observed by the user to ensure safe operation of
the instrument and to retain it in safe condition. Therefore, readthrough these operating instructions before starting using the
instrument.

# WARNING●Read through and understand instructions contained in this manual
before starting to use the instrument.
●Save and keep the manual handy to enable quick reference whenevernecessary.
●Be sure to use the instrument only in its intended applications and tofollow measurement procedures described in the manual.
●Be sure to understand and follow all safety instructions contained in the
manual.
Be sure to observe the above instructions.
Failure to follow the above instructions may cause injury, instrument
damage and/or damage to equipment under test.
Kyoritsu is by no means liable for any damage resulting from the
instrument in contradiction to this cautionary note.
○The symbol # indicated on the instrument means that the user mustrefer to related parts in the manual for safe operation of the instrument.
Be sure to carefully read the instructions following each # symbol inthis manual.

DANGER is reserved for conditions and actions that are likely to#cause serious or fatal injury.

WARNING is reserved for conditions and actions that can cause#
serious or fatal injury.

CAUTION is reserved for conditions and actions that can cause#
minor injury or instrument damage.

—1—

○ Following symbols are used on the instrument and in the instruction
manual. Attention should be paid to each symbol to ensure your safety.
Refer to the instructions in the manual.
Indicates an instrument with double or reinforced insulation.
Indicates that this instrument can clamp on bare conductors whenmeasuring a voltage corresponding to the applicable Measurement
category, which is marked next to this symbol.
Indicates AC (Alternating Current).
Indicates DC (Direct Current).
Indicates AC and DC.
Indicates Earth.

#

DANGER●Never make measurement on the circuit above 750VAC or 1000VDC.
●Do not attempt to make measurement in the presence of flammablegasses, fumes, vapor or dust. Otherwise, the use of the instrument may
cause sparking, which can lead to an explosion.
●Transformer jaw tips are designed not to short the circuit under test. If
equipment under test has exposed conductive parts, however, extra
precaution should be taken to minimize the possibility of shorting.
●Never attempt to use the instrument if its surface or your hand is wet.
●Do not exceed the maximum allowable input of any measurement range.
●Never open the battery compartment cover and the instrument case when
making measurement.
●Verify proper operation on a known source before use or taking action
as a result of the indication of the instrument.
●Never try to make measurement if any abnormal conditions, such as
broken Transformer jaws or case is noted.
●The instrument is to be used only in its intended applications or conditions.
Otherwise, safety functions equipped with the instrument doesn't work, andinstrument damage or serious personal injury may be caused.
●Keep your fingers and hands behind the Barrier and protective fingerguard
during measurement.

WARNING●Never attempt to make any measurement if any abnormal conditions arenoted, such as broken case, cracked test leads and exposed metal parts.
●Do not turn the function selector switch with test leads connected to the
instrument.
●Do not install substitute parts or make any modification to the instrument.
Return the instrument to your distributor for repair or re-calibration.
●Do not try to replace the batteries if the surface of the instrument is wet.
●Always switch off the instrument and make sure to disconnect test leads
before opening the battery compartment cover for battery replacement.
●Stop using the test lead if the outer jacket is damaged and the inner metal orcolor jacket is exposed.

—2—

#CAUTION●Always make sure to check the function selector switch is set to an
appropriate position before starting measurement.
●Always make sure to insert the plug of each lead fully into theappropriate terminal on the instrument.
●Be sure to set the function selector switch to the "OFF" position afteruse. When the instrument will not be in use for a long period, place it instorage after removing the batteries.
●Do not expose the instrument to the direct sun, high temperature or dew
fall.
●Use a cloth dipped in water or neutral detergent for cleaning the
instrument. Do not use abrasives or solvents.
Measurement categories (Over-voltage categories)
To ensure safe operation of measuring instruments, IEC 61010 establishes
safety standards for various electrical environments, categorized as O to
CAT IV, and called measurement categories. Higher-numbered categories
correspond to electrical environments with greater momentary energy, so ameasuring instrument designed for CAT III environments can endure greater
momentary energy than one designed for CAT II.
OCAT IICAT IIICAT IV: Circuits which are not directly connected to the mains power
supply.
: Electrical circuits of equipment connected to an AC electricaloutlet by a power cord.
: Primary electrical circuits of the equipment connected directly tothe distribution panel, and feeders from the distribution panel tooutlets.
: The circuit from the service drop to the service entrance, andto the power meter and primary overcurrent protection device(distribution panel).

O: Device which is
not directly
connected to themains power supply—3—２．Features●Tear-drop-shaped jaws for ease of use in crowded cable areas and othertight places.
●Accurate true-RMS reading of AC current or voltage with distorted
waveform (KEW SNAP 2002R)●Provides a wide measuring range from 0 up to 2000A.
●Terminal cover to avoid the use of an incorrect terminal.
●Measures current variation as short as 10 msec with peak-hold feature.
●Provides output to a chart recorder for current variation recording.
●Designed to international safety standards.
IEC61010-1 (CAT III 600V /CAT II 1000V Pollution degree 2)
IEC61010-031, IEC61010-2-032, IEC 61010-2-033●Data hold function to allow easy readings in dimly lit or hard-to-readlocations.
●Sleep feature to conserve battery power.
●Permits easy continuity check with a beeper.
●Provides a dynamic range of 4,000 counts full scale.
●Provides wide measuring range of voltage and resistance in auto-ranging.
●Wide frequency range from 40Hz to 1kHz (Current measuring range:
0-1500A).
●Transformer jows fitted with guard to further improve safety.
●Protected throughout by double or reinforced insulation  .

—4—

３．Specifications
３−１ KEW SNAP 2002PA●Measuring Ranges and Accuracy(at 23±5℃, relative humidity 45-75%)
AC Current
400Ａ，
2000Ａ

〜

〜

Range

Measuring Range

400Ａ

０〜400.0Ａ

0.1Ａ

2000Ａ

０〜1000Ａ
1000〜1500Ａ
1500〜2000Ａ

１Ａ

Resolution

Accuracy（Frequency Range）±1.0％rdg±3dgt(50/60Hz）±2.0％rdg±3dgt(40〜１kHz)±1.0％rdg±3dgt(50/60Hz）±3.0％rdg±3dgt(40〜１kHz)±3.0％rdg(50/60Hz）〜Ｖ）Auto-ranging

Maximum Measurement Time

Continuous
15min
５min

AC Voltage（

Range
Measuring Range Resolution
Accuracy（Frequency Range）
０〜40.00Ｖ
0.01Ｖ ±1.0％rdg±2dgt(50/60Hz）
40Ｖ
400Ｖ
15.0〜400.0Ｖ
0.1Ｖ±1.5％rdg±3dgt(40〜１kHz)
750Ｖ
150〜750Ｖ
１Ｖ
Initially set to the 40V range. Input impedance is about 1MΩ.

DC Voltage（

Ｖ）Auto-ranging

Range
Measuring Range Resolution
０〜±40.00Ｖ
0.01Ｖ
40Ｖ
400Ｖ
±1.0％rdg±2dgt±15.0〜±400.0Ｖ 0.1Ｖ
1000Ｖ ±150〜±1000Ｖ １Ｖ
Initially set to the 40V range. Input impedance is about 1MΩ.

Accuracy

Resistance(Auto-ranging)

Accuracy
Range
Measuring Range Resolution
０〜400.0Ω
0.1Ω
400Ω
4kΩ
0.150〜4.000kΩ
１Ω
40kΩ
10Ω
±1.5％rdg±2dgt
1.50〜40.00kΩ
400kΩ 15.0〜400.0kΩ
100Ω
Initially set to the 400Ω range. In the continuity check mode, fixed to the 400Ω range andwhen the reading is not more than 50±35Ω, the buzzer beeps.

ＯＵＴＰＵＴ（AC Current Ranges）

DC Output: 100.0mV per 1000 counts (Output impedance: about 10kΩ)Range Output Voltage / Measuring Range
Accuracy（Frequency Range）
400Ａ ０〜400.0ｍＶ/０〜400Ａ±1.5％rdg±0.5ｍＶ(50/60Hz）±2.5％rdg±0.5ｍＶ(40〜１kHz）
2000Ａ ０〜150.0ｍＶ/０〜1500Ａ±1.5％rdg±0.5ｍＶ(50/60Hz）±3.5％rdg±0.5ｍＶ(40〜１kHz）
150.0〜200.0ｍＶ/1500〜2000Ａ ±3.5％rdg(50/60Hz）●Electromagnetic compatibility (EMC)
EN61000-4-3 Radiated RF electromagnetic field immunity
RF field strength＝＜＝１Ｖ／ｍ, total accuracy：specified accuracy
RF field strength＝３Ｖ／ｍ, total accuracy：specified accuracy＋２％ of range—5—

● Operating System● Display● Low Battery Warning● Overrange Indication● Response Time● Sleep function● Data Hold
● Operating Environmental
conditions● Storage Temperatureand Humidity● Operating Temperatureand Humidity● Conductor Size
● Overload Protection● Withstand Voltage● Insulation Resistance● Safety Standard● EMC● Environmental standards● Dimensions● Weight● Power Source● Current Consumption● Accessories● Optional Accessories

Dual Integration
Liquid crystal display with a maximum countof 4000"BATT" symbol is displayed on the digital
display.
"OL" is displayed where input exceeds theupper limit of a range
Approx. 2 seconds
Automatically powered down in about 10
minutes after the last switch operation
Available in all ranges provided the peakmeasurement mode is deactivated.
indoor use
altitude up to 2000m
ー 20 〜 60℃ , relative humidity up to 85%
without condensation
0 〜 40℃ , relative humidity up to 85%
without condensation
Approx. 54.5 diameter max.
2400A AC for 10sec
1200V AC/DC for 10sec
600V AC for 10sec
5160V AC for 5 seconds between electricalcircuit and housing cases or metal parts ofjaws
10M Ω or greater at 1000V between
electrical circuit and housing cases or metalparts of jaws
IEC 61010-1, 61010-2-032, 61010-2-033,
61010-031: Measurement CAT Ⅲ 600V/
CAT Ⅱ 1000V, pollution degree 2.
EN 61326-1
EU RoHS directive compliant
247（L）× 105（W）× 49（D）mm
Approx. 470g(battery included)
Two R6P(DC1.5V) batteries or equivalent
Approx. 5mA max. (Approx. 20 μ A in thesleep mode)
Test leads M-7107A
Two R6P batteries
Instruction manual
Carrying case M-9094
Output Probe M-7256, etc.

—6—３−２ KEW SNAP 2002R● Measuring Ranges and Accuracy(at 23 ± 5℃ , relative humidity 45-75%)
AC Current
400Ａ，
2000Ａ
(9 counts or less is corrected to 0)
Range

Measuring Range〜〜
Resolution

400Ａ

０〜400.0Ａ

0.1Ａ

2000Ａ

０〜1000Ａ
1000〜1500Ａ
1500〜2000Ａ

１Ａ

Accuracy（Frequency Range）（45〜65Hz）±1.5%rdg±3dgt（40〜1kHz）±2.5%rdg±3dgt（45〜65Hz）±2.0%rdg±5dgt（40〜1kHz）±3.0%rdg±5dgt（50/60Hz）±4.0%rdg

Maximum Measurement Time

Continuous
15min
５min

〜

AC Voltage（

Ｖ）Auto-ranging(9 counts or less is corrected to 0)
Range
Measuring Range Resolution
Accuracy（Frequency Range）
０〜40.00Ｖ
0.01Ｖ ±1.0％rdg±2dgt(45〜65Hz)
40Ｖ
400Ｖ
15.0〜400.0Ｖ
0.1Ｖ±1.5％rdg±3dgt(40〜１kHz)
750Ｖ
150〜750Ｖ
１Ｖ
Initially set to the 40V range. Input impedance is about 1M Ω .
When approx.300V or more is applied to the instrument instantaneously, the measured value
is indicated on 750V range.

DC Voltage（

Ｖ）Auto-ranging

Accuracy
Range
Measuring Range Resolution
０〜±40.00Ｖ
0.01Ｖ
40Ｖ
400Ｖ
0.1Ｖ
±1.0％rdg±2dgt±15.0〜±400.0Ｖ
1000Ｖ ±150〜±1000Ｖ
１Ｖ
Initially set to the 40V range. Input impedance is about 1M Ω .

Resistance(Auto-ranging)

Range
Measuring Range Resolution
Accuracy
０〜400.0Ω
0.1Ω
400Ω
4kΩ
0.150〜4.000kΩ
１Ω±1.5％rdg±2dgt
40kΩ
10Ω
1.50〜40.00kΩ
400kΩ 15.0〜400.0kΩ
100Ω
Initially set to the 400 Ω range. In the continuity check mode, fixed to the 400 Ω range andwhen the reading is not more than 50 ± 35 Ω , the buzzer beeps.

ＯＵＴＰＵＴ（AC Current Ranges）

DC Output: 100.0mV per 1000 counts (Output impedance: about 10k Ω )Accuracy
Range Output Voltage / Measuring Range（Frequency Range）（45〜65Hz）
400Ａ ０〜400.0ｍＶ/０〜400Ａ±2.0%rdg±0.5mV（40〜1kHz）±3.0%rdg±0.5mV
2000Ａ ０〜150.0ｍＶ/０〜1500Ａ（45〜65Hz）±2.5%rdg±0.5mV（40〜1kHz）±3.5%rdg±0.5mV
150.0〜200.0ｍＶ/1500〜2000Ａ（50/60Hz）±4.5%rdg● CF (Crest Factor) CF=3 or lessaccuracy+1% (45 〜 65Hz)、less than AC3000A/AC1200V Peak
● Electromagnetic compatibility (EMC)
EN61000-4-2 Electrostatic discharge immunity(ESD)
Performance criteria B—7—

● Operating System● Display● Low Battery Warning● Overrange Indication● Response Time● Sleep function● Data Hold
● Operating Environmental
conditions● Storage Temperatureand Humidity● Operating Temperatureand Humidity● Conductor Size
● Overload Protection● Withstand Voltage● Insulation Resistance● Safety Standard● EMC● Environmental standards● Dimensions● Weight● Power Source● Current Consumption● Accessories● Optional Accessories

Dual Integration
Liquid crystal display with a maximum countof 4000"BATT" symbol is displayed on the digital
display.
"OL" is displayed where input exceeds theupper limit of a range
Approx. 2 seconds (at full scale)
Automatically powered down in about 10
minutes after the last switch operation
Available in all ranges provided the peakmeasurement mode is deactivated.
indoor use
altitude up to 2000m
ー 20 〜 60℃ , relative humidity up to 85%
without condensation
0 〜 40℃ , relative humidity up to 85%
without condensation
Approx. 54.5 diameter max.
2400A AC for 10sec
1200V AC/DC for 10sec
600V AC for 10sec
5160V AC for 5 seconds between electricalcircuit and housing cases or metal parts ofjaws
50M Ω or greater at 1000V between
electrical circuit and housing cases or metalparts of jaws
IEC 61010-1, 61010-2-032, 61010-2-033,
61010-031: Measurement CAT Ⅲ 600V/
CAT Ⅱ 1000V, pollution degree 2.
EN 61326-1
EU RoHS directive compliant
247（L）× 105（W）× 49（D）mm
Approx. 470g(battery included)
Two R6P(DC1.5V) batteries or equivalent
Approx. 10mA max. (Approx. 20μ A in thesleep mode)
Test leads M-7107A
Two R6P batteries
Instruction manual
Carrying case M-9094
Output Probe M-7256, etc.

—8—

—9—

４．Instrument Layout

Transformer Jaw
BarrierProtective
fingaguard

Jaw Trigger

Function Switch
Data Hold Switch
Display
Reset Switch︸

Terminal CoverMode Switch

Drop Prevention Strap
OUTPUT Terminal
COM Terminal

Test Leads
M-7107A
Ｖ／Ω Terminal

Barrier and Protectivefingerguard:
It is a part providing protection against electrical shock and ensuring the minimum required
air and creepage distances.
Test Lead Cap：Test leads can be used under the CAT II and III environments
by attaching a Protective cap as illustrated below. Use of our Protective
cap offers different lengths suitable for the test environments.
Protective Cap

Exposed metal part

Uncapped condition for CAT II environment

Capped condition for CAT III environments

When the instrument and the test lead are combined and used together, whichever lower category
either of them belongs to will be applied.

— 10 —

● ＬＣＤ INDICATORDC Current

AC Current

Data Hold Mode
Resistance

Negative Potential

Voltage
Current

Continuity Check Mode

Battery Voltage Alert

Peak Mode Measurement

５．Preparation for Measurement
５−１ Checking Battery Voltage① Set the function selector switch to any position other than "OFF".
② When the display is clear without "BATT" showing, proceed tomeasurement.
③ When the display blanks or "BATT" is indicated, replace the batteries
according to section 8: battery replacement.
NOTE● It is possible that display is kept in blank while the function selector
switch is set to a position other than "OFF". This is due to sleep
function which automatically powers the instrument down in a certain
period of time after the last switch operation. To operate the instrumentin this case, set the switch back to the "OFF" position, then to thedesired position, or press any button.

５−２

Checking Switch Setting and Operation

Make sure that the function selector switch is set to the correct position,
the instrument is set to the correct mode and the data hold function isdeactivated. Otherwise, desired measurement cannot be made. (See section
6 for measurement instructions and section 7 for notes on functions.)— 11 —

６．Measurement
６−１ Current Measurement

WARNING● Do not make measurement on a circuit above 750V AC. This may
cause shock hazard or damage to the instrument or equipment undertest.
● Transformer jaw tips are designed not to short the circuit under test. If
equipment under test has exposed conductive parts, however, extra
precaution should be taken to minimize the possibility of shorting.
● Do not make measurement with the battery compartment cover removed
from the instrument.
● Do not make current measurement with the test leads connected to theV/ Ω and COM terminals.
● When measuring current is not less than 1000A, make sure to stopmeasurement within the maximum measuring time shown below.
Otherwise, transformer jaws may heat to cause a fire or deformation ofmolded parts, which will degrade insulation.
1000 〜 1500 Ａ：15min. 1500 〜 2000 Ａ：５min.
● Keep your fingers and hands behind the barrier during measurement.

Incorrect

Correct

６−１−１ AC Current Measurement (Normal Mode)① Set the function selector switch to the " 〜 400A" or " 〜 2000A"
position and make sure that the current under test does not exceed theupper limit of the measuring range you are selecting.
② Press the trigger to open the transformer jaws and clamp them onto the
conductor under test.
— 12 —

③ Take the reading on the display.

NOTE● During current measurement, keep the transformer jaws fully closed.
Otherwise, accurate measurement cannot be made. The maximum
measurable conductor size is 54.5mm in diameter.
● When measuring a larger current, the transformer jaws may buzz. This
is not a fault and does not affect the accuracy at all.

６−１−２

Peak Current Measurement① Set the function selector switch to the " 〜 400A" or " 〜 2000A" position.
② Press the mode switch to select the peak mode. "PEAK" will be shownon the display.
③ Press the trigger to open the transformer jaws and clamp them onto the
conductor under test. Then, press the reset switch.
④ The display shows the current's crest value divided by the square rootof two. Therefore, when the current is sinusoidal, the reading equals
RMS value.
⑤ To reset the display, press the reset switch.
（Note: When this is done, the reading goes off for about one second.）

INPUT Current

Peak Hold
Ｖｐ

Ｖｐ／√2

⑥ After the measurement is over, press the mode switch to return to thenormal mode.

NOTE● In the peak measurement mode, the data hold feature is disabled.
● When a measured value is 9 counts or less, it is corrected to 0 (KEW
SNAP 2002R).
— 13 —

６−２ Voltage Measurement

WARNING●Do not make measurement on a circuit above 750V AC or 1000V DC.
This may cause shock hazard or damage to the instrument or equipment
under test.
●Do not make measurement with the battery compartment cover removed.
●Keep your fingers and hands behind the protective fingerguard during
measurement.

Black test lead

Red test lead

６−２−１

DC Voltage Measurement

６−２−２

AC Voltage Measurement① Set the function selector switch to the "V" position.
② Slide the terminal cover to the left. Plug the red test lead into the V/ Ω
terminal and the black test lead into the COM terminal.
③ Connect the other end of the test leads to the circuit under test. Take
the reading on the display. When the red lead is the negative potential,
the " ー " sign is shown on the display.
① Set the function selector switch to the " 〜 V" position.
② Slide the terminal cover to the left. Plug the red test lead into the V/ Ω
terminal and the black test lead into the COM terminal.
③ Connect the other end of the test leads to the circuit under test. Take
the reading on the display.

NOTE● For high sensitivity, there are parts which do not indicate " ０".

— 14 —

６−３

Resistance Measurement

WARNING●Before attempting to make measurement, make sure that the circuitunder test is not live. The instrument is protected against a voltage up
to 600V.
●Do not make measurement with the battery compartment cover removed.
●Keep your fingers and hands behind the protective fingerguard during
measurement.

Black
test lead

６−３−１

Red test lead

Resistance Measurement (Normal Mode)① Set the function selector switch to the " Ω / " position.
② Slide the terminal cover to the left. Plug the red test lead into the V/ Ω
terminal and the black test lead into the COM terminal.
③ Short the tip of the test leads and check whether the display reads "0".
④ Connect the tip of the test leads to the circuit under test. Take thereading on the display.

NOTE● When shorting the tip of the test leads, the display may read a verysmall resistance instead of "0." This is the resistance of the test leads,
not a fault.
● If one of the test leads is open, the display reads "OL".
— 15 —

６−３−２

Continuity Check① Set the function selector switch to the " Ω /" position.
② Slide the terminal cover to the left. Plug the red test lead into theV/ Ω terminal and the black test lead into the COM terminal.
③ Press the mode switch to set the instrument to the continuity checkmode. The measuring range is fixed to 400 Ω and the " " symbol
is shown on the display.
④ Short the tip of the test leads and make sure that the display reads"0" and the buzzer beeps.
⑤ Connect the tip of the test leads to the circuit under test. The
display reads the resistance and the buzzer beeps when thereading is not more than about 50 Ω .

NOTE● When shorting the tip of the test leads, the display may read a verysmall resistance instead of "0". This is the resistance of the test
leads, not a fault.
● If one of the test leads is open, the display reads "OL".

— 16 —

７．Notes on Functions
７−１ Data HoldThis is a function used to freeze the measured value on thedisplay.
① Press the data hold switch. The reading becomes frozen and the"H" symbol is shown on the display, indicating the instrument in
the data hold mode.
② To exit the data hold mode, press the data hold switch again torelease it.

NOTE● When the function selector switch is turned while the instrument isin the data hold mode, the data hold function remains activated.
To make measurement in this case, release the data hold switch
by pressing it and exit the data hold mode.
● The data hold function is disabled in the peak measurement modeon the AC current range.
● When the sleep function is activated, the data hold mode turns to
the normal mode.

７−２

Sleep Function

This is a function to prevent the instrument from being left powered
on in order to conserve battery life.
① The instrument automatically enters the sleep (powered-down)
mode about 10 minutes after the last switch operation.
② To exit the sleep mode, press the data hold, reset or mode switch
or turn the function selector switch back to "OFF", then to any
other position.

［How to Exit the Sleep Mode ］① Turn the function selector switch from "OFF" to another position
with the data hold switch pressed. Then, "P.OFF" is shown on thedisplay. This disables the sleep function and enables continuous
use of the instrument.
— 17 —

② To enable the sleep function, turn the function selector switch backto "OFF", then to any other position.

NOTE● The instrument consumes small amount of current in the sleep
mode. When the instrument is not in use, make sure to set thefunction selector switch to "OFF".

７−３

Recorder Output

Only on the " 〜 400A" or " 〜 2000A" range, DC voltage proportional
to the input current is output from the OUTPUT terminal.
① Set the function selector switch to the " 〜 400A" or " 〜 2000A"
position.
② Slide the terminal cover to the right and insert the recorder output
plug into the OUTPUT terminal for connection with a recorder orother recording device.

NOTE● Output voltage is 1mV/A on the " 〜 400A" range and 0.1mV/A on
the " 〜 2000A" range. Set an appropriate input sensitivity on therecorder.
● The peak hold function does not apply to the recorder output even if
the instrument is in the peak hold mode.
● For long term measurement, disable the sleep function. (Seesection 7-2 for sleep function.)— 18 —DANGER

Never apply voltage to the output terminal.

７−４Mode Switching Function

On a AC current (" 〜 400A" or " 〜 2000A") or the resistance( Ω /)range, press the mode switch to cycle through the measurement modes.
The instrument is initially set to the normal mode and can be switched tothe peak or continuity check mode by means of the mode switch. (Seesection 6-1-2 for peak current measurement and section 6-3-2 forcontinuity check.)≪ AC Current Range(400A or 2000A) ≫NormalPress the mode switch

Press the mode switch

Peak
Peak current value is displayed.
(Response time: 10ms)"PEAK" is shown on the display.

≪ Resistance Range ≫NormalPress the mode switch

Press the mode switch

Continuity Check
The buzzer beeps if the resistance is
not more than about 50 Ω .
The buzzer symbol is shown on thedisplay.

— 19 —

８．Battery Replacement
WARNING● To avoid electric shock hazard, make sure to set the function selector
switch to "OFF" and remove the test leads from the instrument before
trying to replace batteries.

CAUTION● Do not mix new and old batteries.
● Make sure to install batteries in correct polarity as indicated in thebattery compartment.

NOTE● If the instrument is powered on, but the display blanks or "BATT" isshown on the display, replace the batteries.
① Set the function selector switch to the "OFF" position.
② Unscrew and remove the battery compartment cover on the bottom of
the instrument.
③ Replace the batteries observing correct polarity. Use two new R6P
batteries.
④ Replace and screw the battery compartment cover.
Screw

Battery
compartment
cover

Battery— 20 —

９．Disposing the ProductWaste Electrical and Electronic Equipment (WEEE), Directive 2002/96/EC
This Product complies with the WEEE Directive (2002/96/EC) marking
requirement.
The affixed product label (see below) indicates that you
must not discard this electrical/electronic product indomestic household waste.
Product Category
With reference to the equipment types in the WEEEd i r e c t i v e A n n e x 1, t h i s p r o d u c t i s c l a s s i f i e d a s
a Monitoring and Control instrumentation product.

— 21 —

DISTRIBUTOR

Kyoritsu reserves the rights to change specifications ordesigns described in this manual without notice and withoutobligations.

8-22

92-1579K