Standard-Level Precision Balances

FZ-i / FX-iSeries

FZ-iFX-i

Pay for What Matters—Reliable,
Accurate Weighing

www.aandd.jp

High Performance and Functionality
You Can Aﬀord
Built on A&D's cutting-edge sensor technology, the FZ-i/FX-i series is ahighly robust precision balance that delivers fast and accurate weighing
without any frills, oﬀering cost-eﬀective solutions for a diverse array ofapplications.

Air-pump controlled internal calibration◆(FZ-i models only)◆ Patented

For the internal calibration of the FZ-i series, A&D adopted a direct-driving
method that utilizes air pressure for its power source. This innovative methodfeatures a simple structure, high reliability and a fail-safe mechanism—theinternal mass returns to the original position even if the calibration isinterrupted by power failure, etc.

One-touch automatic calibration (FZ-i models only)
You can ensure the accuracy of the balance with just one key press. No special
skills are required. This is useful after you changed the location of the balanceor when there was a ﬂuctuation in environment, etc.

Compact footprint: 198 mm × 294 mm
By taking advantage of our Compact Super Hybrid Sensor (C-SHS) technology◆, A&D signiﬁcantly
reduced the size of a precision balance—25% smaller footprint than our conventional models,
making it ideal for installation and use in conﬁned spaces.
◆ Patented

High-speed weighing with 1-second stabilization
The C-SHS combines the best of magnetic force restoration and single point parallelogram load sensing
technologies. This allows the C-SHS to realize a stabilization speed of just 1 second while maintaining
extremely high span stability. The FZ-i/FX-i series will maximize your productivity and eﬃciency.

Guarded against accidental overloading ◆An internal shock absorber*1 protects the C-SHS from the adverse eﬀects of vertical and transverse
overloading. This is especially eﬀective if the balance is embedded in an automated weighing system
where actuator malfunctions are a possibility.
◆ Patented*1 Not designed to handle impact loads

RS-232C interface (standard)
Bi-directional communication is possible with a PC, printer or other peripheral devices.

Quick USB interface (FXi-02)
The easy-to-use optional USB interface requires no driver or software installation for transfer ofweighing data to a PC.*2
2 Data transfer from the balance to a PC only. Please use the RS-232C standard* interface with the serial/USB converter (AX-USB-9P, optional) for two-way
communication or sending GLP or statistical calculation information via USB.
Portable memory devices such as weighing environment logger (AD-1687) and datalogger (AD-1688) are also available.

Ethernet interface (FXi-08) with WinCT-Plus software
You can send commands for controlling and acquiring data from multiple balances.

WinCT-PlusLAN

Ethernet HubUSB

Provided by end-user

FXi-08FXi-08FXi-08FXi-02Multiple units of measure
You can easily select from: g, oz, lb, lb-oz, ozt, ct, mom, dwt, gr, pcs (counting mode), % (percent mode)
and a user-programmable unit for conversion applications. One additional unit from tl, tol, or N can beadded upon request.

GLP/GMP/GCP/ISO compliant output
For documentation requirements, the balance can output its manufacturer, model, serial number, IDnumber, date + time*3, space for signature for calibration report, calibration test report, and title & end
blocks for a series of weighing results.
3 For the FZ-i models only. Please use the AD-8127 compact printer, which* has a calendar function, to print the date and time with the FX-i models.

Statistical calculation function (SCF)
The balance can display and output statistical calculation data including number of data, sum,maximum, minimum, range (maximum–minimum), average, standard deviation, and coeﬃcient ofvariation to facilitate the analysis of measurements.

Adjustable response characteristics and holdfunction
From three preset combinations of weighing speed and stability(FAST, MID, or SLOW), select the one that best suits yourenvironmental conditions (drafts and vibrations). A holdfunction is also available for weighing animals.

Automatic counting accuracy improvement (ACAI)
For the counting function, the balance recalculates the average
unit weight as samples are added to eliminate errors caused byvariations in unit weight.

Built-in rechargeable battery (FXi-09)
The Ni-MH rechargeable battery pack can be ﬁtted into thebalance for use without AC power (10 hours of charging for 8hours of operation).

Large Breeze Break (FXi-11)

A small breeze break is provided as
standard for all the FZ-i models,FX-120i, FX-200i and FX-300i.

Speciﬁcations
Capacity

FZ -120i
FX-120i

FZ -200i
FX-200i

FZ-300i
FX-300i

FZ-500i
FX-500i

FZ-1200i
FX-1200i

FZ-2000i
FX-2000i

FZ-3000i
FX-3000i

FZ-5000i
FX-5000i

122 g

220 g

320 g

520 g

1220 g

2200 g3200 g5200 g0.001 gReadability0.01 g

Repeatability (standard deviation)

0.001 g0.01 gLinearity±0.002 g±0.02 gStabilization time
(when set to FAST under a good environment)

Approx. 1 second

Sensitivity drift±2 ppm /°C (10 to 30 °C/50 to 86 °F)

Operating environment

5 to 40 °C (41 to 104 °F), 85% RH or less (no condensation)

Display type

Vacuum Fluorescent Display (VFD)

Display refresh rate

5 times /second, 10 times /second or 20 times /second

Units of measure*ig (gram), oz (ounce), lb (pound), lb-oz (pound-ounce), ozt (troy ounce), ct (metric carat), mom (momme),
dwt (pennyweight), gr (grain), pcs (counting mode), % (percent mode), and a user-programmable unit

Counting mode Minimum unit mass

0.001 g0.01 g5, 10, 25, 50 or 100 pieces

Number of samples
Percent mode

0.100 gMinimum 100% reference mass

1.00 g0.01 %, 0.1 %, 1 % (depends on the reference mass stored)% readability
Standard interface

RS-232C

Weighing pan size (diameter)

130 mmInternal calibration

150 mmAir-pump controlled internal calibration (FZ-i models only)

Applicable calibration weight value

100 g
50 g

AC adapter

200 g
100 g
50 g

300 g
200 g
100 g

500 g
400 g
300 g

50 g

200 g

1000 g500 g2000 g1000 g500 g3000 g2000 g1000 g5000 g4000 g3000 g500 g2000 g100 g1000 g

50 g

500 g

Conﬁrm that the adaptor type is correct for the local voltage and power receptacle

Power consumption

Approx. 11VA (supplied to the AC adaptor)

External dimensions

193 (W) × 262.5 (D) × 85.5 (H) mm

Net weight

FZ-i Approx. 3.0 kg / FX-i Approx. 2.5 kg*i One additional unit from tael (Singapore/HK jewelry/Taiwan/China), tola or Newton can be added upon request.
Options
FX i-02*ii
FX i-08*ii
FX i-09*ii
FX i-10

Quick USB interface with cable
Ethernet interface with WinCT-Plus software
Built-in rechargeable battery
Small breeze break

AD-1683
AD-1684A
AD-1687
AD-1688
AD-1689
AD-8920A
AD-8922A
AD-8127

Static eliminator
Electrostatic ﬁeldmeter
Weighing environment logger
Weighing data logger
Tweezers for calibration weight
Remote display
Remote controller
Compact printerAccessories

FX i-11 Large breeze breakFX i-12 Animal weighing pan (for the 1200/2000/3000/5000 g capacity models only)
FX i-15 Carrying case*ii FXi-02, FXi-08 and FXi-09 cannot be used at the same time.

Serial/Ethernet converter
AD-8526
Quick USB adapter
AD-8527
Serial/USB converter with cable
AX-USB-9P
AX-KO2466-200 RS-232C cable (9P-9P)
AD-8529PC-W Bluetooth converter for PC
AD-8529PR-W Bluetooth converter for printer
Main unit cover (5 pcs)
AX-FX i-31

AD-1687

AD-168812936Internal
Calibration

External
Calibration

FZ-i

FX-i

Comparator
Function

Auto PowerON

Super Hybrid
Sensor

RS-232C
Interface

GLP
Compliant

Clock & Calendar

Countingwith ACAI

Percentage
Function

AnimalWeighingStatistical
Calculation

Optional

Quick USB

LAN-Ethernet
InterfaceRechargeable
BatteryFZ-iAuto PowerOFF

SoftwareUnderhook

FZ-i

A&D Company, Ltd.

3-23-14 Higashi-Ikebukuro, Toshima-Ku, Tokyo, 170-0013, Japan Tel: +81 3-5391-6132 Fax: +81 3-5391-1566 www.aandd.jp

A&D Engineering, Inc.

1756 Automation Parkway, San Jose, CA 95131, U.S.A. Tel: +1 408-263-5333 Fax: +1 408-263-0119

A&D Australasia Pty Ltd.

32 Dew Street, Thebarton, South Australia 5031, Australia Tel: +61 8-8301-8100 Fax: +61 8-8352-7409

A&D Instruments Ltd.

Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire, OX14 1DY, United Kingdom Tel: +44 1235-550420 Fax: +44 1235-550485<German Sales Office>Hamburger Straße 30, D-22926, Ahrensburg, Germany Tel: +49 4102-459230 Fax: +49 4102-459231

A&D Korea Ltd.

8F Manhattan Bldg., 33, Gukjegeumyung-ro 6-gil, Yeongdeungpo-gu, Seoul, 07331, Korea Tel: +82 2-780-4101 Fax: +82 2-782-4280

A&D Rus Co., Ltd.

Vereyskaya Str. 17, 121357, Moscow, Russia Tel: +7 495-937-33-44 Fax: +7 495-937-55-66

A&D Instruments India (P) Ltd.

509 Udyog Vihar Phase V Gurgaon-122 016, Haryana, India Tel: +91 (124) 471-5555 Fax: +91 (124) 471-5599* FZi/FXi-ADCC-08-PR8-19501GP