Product Information

IBV 600 Series
Interpolation andDigitizing Electronics

September 2010

IBV 600 Series
Interpolation and digitizing electronics• Input » 1 VPP• Output « TTL

mmTolerancing ISO 8015
ISO 2768 - m H< 6 mm: ±0.2 mm

57175
¬ 4.3

80

52±0.240¬ 8.5

M4 x 20 ISO 4762

23HEIDENHAIN

M4



2274

163±0.2

614599ÀProduct Information for IBV 600 Series

9/2010Specifications

IBV 600

Input» 1 VPPElectrical connection

M23 flange socket (female) 12-pin

Cable length† 60 m at UP > 4.9 V† 30 m at Iencoder † 120 mA1)IBV 660 BInterpolationNone

25-fold, 50-fold, 100-fold, 200-fold, 400-fold

Input frequency1) forinterpolation

Nominal values2)

None 600 kHz–2-fold ––25-fold –100 kHz

50 kHz

25 kHz12.5 kHz

50-fold –50 kHz

25 kHz12.5 kHz6.25 kHz

100-fold –25 kHz12.5 kHz6.25 kHz3.12 kHz200-fold –12.5 kHz6.25 kHz3.12 kHz

1.56 kHz

400-fold –6.25 kHz3.12 kHz

1.56 kHz

0.78 kHz‡ 0.175 µs‡ 0.370 µs‡ 0.760 µs

Output1« TTL (nonclocked)

Electrical connection

M23 flange socket (male) 12-pin

Cable length† 100 m (¥ † 50 m)

Edge separation a‡ 0.200 µs1)

« TTL (clocked)‡ 0.075 µs

Reference mark signal

Pulse width 90° elec. or 270° elec.

Fault indication1)

through fault detection signal ¥ or, in addition, Ua1/Ua2 high impedancePower supply

5 V ± 5%3)

Current consumption† 85 mA† 120 mA

Operating temperature
Storage temperature

0 °C to 70 °C–30 °C to 80 °C

Vibration 50 to 2 000 Hz
Shock 11 ms† 10 m/s2† 300 m/s2

Degree of protection

IP 65

WeightApprox. 0.7 kg

Bold: This version is the factory default setting
1)
Adjustable2)
The actual input frequency can be up to 5 % lower. Exceeding this limit results in failure3)
Not including the current consumption of the encoder (see encoder brochure) and without output load
(80 mA with recommended input circuitry)

Product Information for IBV 600 Series

9/2010Electrical Connection

Connecting cable or adapter
cable with M23 connector(male) 12-pin

Connecting cable M23 12-pin, ¬ 8 mm

Complete
ID 298 399-xx
With one connector
ID 309 777-xx

Cable and connector, 12-pin.
See also HEIDENHAIN catalogs
for linear encoders, angle
encoders and rotary encoders
as well as Product Information
sheets for the respectiveencoders.

Cable only
ID 244 957-01[4(2x0.14mm2) + (4x0.5 mm2)]

Connector (female), 12-pin
ID 291 697-05

IBV input – » 1 VPP
12-pin flange socket M23

91281012113
4765

Power supply

Incremental signals

Other signals

122

10

115

68

134

7

9UP

Sensor
UP0V

Sensor
0V

A+

A–

B+

B–

R+

R–

Vacant Vacant Vacant

Brown/
GreenBlueWhite/
GreenWhiteBrown

GreenGrayPink

RedBlack

Violet//Yellow

IBV output – « TTL
12-pin flange socket M23

Mating connector:
12-pinconnector
M23Power supply

Incremental signals

12210115UPSensor
5V0VSensor
0V

Ua1Brown/
GreenBlueWhite/
GreenWhiteBrown

6GreenOther signals

81

3

4

7

Ua2

£

Ua0¤

¥

Gray

Pink

RedBlack

Violet/9

Vacant Vacant/Yellow

Cable shield connected to housing; UP = power supply voltage
Sensor: The sensor line is connected in the encoder with the corresponding power line.
Vacant pins or wires must not be used!

For more information• Product overview: Interface Electronics
DR. JOHANNES HEIDENHAIN GmbH
Dr.-Johannes-Heidenhain-Straße 583301 Traunreut, Germany{ +49 8669 31-0
| +49 8669 5061
E-mail: info@heidenhain.dewww.heidenhain.de
598 011 · 01 · A · 02 · 9/2010 · PDF