Infrared Ultrathin Thickness MeterIRMT01
Measure online / continuously the ultra thin film under 10μm!

Infrared Ultra thin Thickness Meter is a contact /onlinemodelthickness

meterthatcancontinuous / high sensitivity measure the film
and coating paint under 10μm.
.

 FEATURES
 Adopt P polarized light and high sensitivity optics

Detector

By eliminating the surface reflectance, it realizes the highaccuracy / high sensitivity measurement.
 Output updating cycle 28ms, high-speed measurement
Real time measurement on the production line.
 Extend the measuring distanceSet the measuring distance as 50mm (our current model25mm) to reduce the effect of the fluctuating measuring
distance.
 Load self-diagnostic function
Output self-diagnostic and easy maintenance.

Setting display unit Prepare Setting display unit and Software package
RS485 is standard equipped. Communicates with the setting
display unit and digital display the measured value, operates
various setting, multi-calculation function and alarm.
Software Package for the PC is also optionally prepared.
 Application

 Models Film Detector

 Measuring the thickness of ultrathin film.

IRMT01

 Measuring the thickness of solvent coating / resin adhesion
 Metal Setting Display Unit
IRGMEG3

 Measuring the thickness of organic film coating/ oiling on themetal plate.

Communication Interface
R : RS232C (standard) Measuring the thickness of corrosion film / insulating film on
the steel plate.

A : RS422A
S : RS485

1PSE-604B

 Configuration

 Software package IR-WEF2

IRGMEG3
RS485

 Main screen
Measured data digital display, trend display, alarm
display, status display.

 Setup screen
Set the selected variety of upper and lower limit/decimal
points/ calibration data register/switch/ communication
setting.

 Calibration curve screen*Used when measuring film

Creates the calibration curve and check the calibrationdata.

 General Specifications
 Detector

 Setting display

Model:
Measuring system:
Light source:
Measuring area:

Model:
Input signal :Output signal:

IRMT01
Infrared ray 3 wavelengths, P polarized light
Tungsten lamp
Approx. 12 x 20 mm (at measuring distance 50mm)
Communications:
RS485 (MODBUS)
Output renewal cycle: 28ms
Working temperature: 0 to 40°C
Power supply:
24V DCPower consumption:
Max. 33VA
Connection:
Power supply, single-line: terminal connection
Casing:Aluminum casting
Weight:Approx. 7kg
Mounting:
Bolt suspension method with 4 pieces of M8bolts
Protection structure:
Dust proof, water proof (conform to IP65)

IRGMEG3□
RS485
Analog output:
4 to 20mA DC, 2output, Load resistance lessthan 500Ω
Communication output:
Select one from RS232C, RS422A or RS485
Display:
Measured data (LED 5-digit), Detector No., CH
No., Parameter setup
Working temperature: 0 to 50°C
Power supply:
100 to 240V AC 50/60HZPower consumption:Max. 15VAConnection:
Power supply, signal-line: terminal connection
Weight:Approx. 0.6 kg
CE marking:
IRGMEG3R (RS232C output model) only
EMC directive --- EN61326-1 Class A
Low voltage directive --- EN61010-1,
EN61010-2-030
RoHS directive --- EN50581

 External dimensions
Detector

Setting display unit

When terminal cover added

Unit: mm

N: No of units

Unit: mm
Specifications subject to change without notice. Printed in Japan. 2020. 1132-8, KUMANO-CHO, ITABASHI-KU, TOKYO 173-8632
PHONE: +81-3-3956-2171
FAX: +81-3-3956-0915E-mail: inter@chino.co.jp

Website: http://www.chino.co.jp

2