SW-D
Weighing Indicator

For Super Washdown Scales

1WMPD4002414A

Warning Definition
The warning described in this manual has the following meaning:

DANGER

An imminently hazardous situation which, if not avoided, will result
in death or serious injury.

© 2011 A&D Company, Limited. All rights reserved.
No part of this publication may be reproduced, transmitted, transcribed, or translated into any
language in any form by any means without the written permission of A&D Company, Limited.

The contents of this manual and the specifications of the instrument covered by this manual are
subject to change for improvement without notice.

CONTENTS
1. COMPLIANCE............................................................................................................... 4
2. INTRODUCTION ........................................................................................................... 5
3. FEATURES.................................................................................................................... 5
4. UNPACKING ................................................................................................................. 6
5. INSTALLATION and PRECAUTIONS........................................................................... 7
5.1. Precautions for Installing the Indicator................................................................ 7
5.2. Connecting to the load cell ................................................................................. 7
5-3. Verifying Load Cell Output and Input Sensitivity................................................. 8
5-4. Setting Up the Indicator ...................................................................................... 7
5-5. Parameter Settings and Calibration.................................................................... 9
6. DESCRIPTION OF EACH PART................................................................................. 10
7. CALIBRATION ............................................................................................................ 127.1. Items of the calibration mode ........................................................................... 127.2. Setting of the weighing capacity / division / decimal point position ................... 127.3. Gravity acceleration correction ......................................................................... 15
7.4. Calibration using a weight ................................................................................ 16
8. SETTING FUNCTIONS OR UNITS FOR USE ............................................................ 18
8.1. Setting of the CF function ................................................................................. 18
8.2. Setting unit for use ........................................................................................... 19
8.3. CF Function list ................................................................................................ 20
9. OPTIONS .................................................................................................................... 21
9.1. SW-11 Display stand / Wall-mount ................................................................... 21
10. MAINTENANCE ........................................................................................................ 2210.1. Notes on maintenance.................................................................................... 2210.2. Error codes ..................................................................................................... 22
11. SPECIFICATIONS ..................................................................................................... 23
11.1. Specifications.................................................................................................. 23
11.2. External dimensions ....................................................................................... 24
GRAVITY ACCELERATION MAP ................................................................................... 25
3

1. COMPLIANCE
Compliance with FCC rules
 Please note that this equipment generates, uses and can radiate radio frequency
energy. This equipment has been tested and has been found to comply with thelimits of Class A digital devices pursuant to Part 15 of FCC rules. These rules aredesigned to provide reasonable protection against interference when equipment is
operated in a commercial environment. If this unit is operated in a residential area, it
may cause some interference and under these circumstances the user would be
required to take, at his own expense, whatever measures are necessary toeliminate the interference. (FCC = Federal Communications Commission in the
U.S.A.)Classification of protection provided by enclosures
 The equipment is designed to comply with the IP Code of IEC 60529.
The “IP69K” is explained as follows:“IP”

International Protection.

“6”

Against ingress of solid foreign objects.
Dust-tight. No ingress of dust.

“9K”

Against ingress of water with harmful effects.
High pressure water jets directed against the enclosure from any direction
shall have no harmful effects. (DIN40050 Part 9)4

2. INTRODUCTION
This manual describes how this product works and how to get the most out of it in termsof performance.
The SW-D weighing indicator may be used to build a water-proof digital scale tospecification IP69K. It has similar specifications and functions as an indicator of the SW
series.
Please refer to the instruction manual for the SW supplied with the SW-D.

3. FEATURESThe SW-D weighing indicator is an electronic display instrument used to build a waterproof scale to specification IP69K by the combination of the SW-D and a weighing unit
(Load cell), with the following features: Dust-tight and water-tight construction, complying with IP69K.
 Employing touch-sensitive keys enabled the display to be covered with a plastic
sheet, resulting in better dust-tight and water-tight performance.
When a key is touched, the corresponding LED, above that key, turns on toindicate that the key has been touched.
 A bright LED display, with a broad viewing angle.
 Operating up to four 350 Ω load cells is available.
 Available range selection, for the display resolution, is 1/20 to 1/60,000.
 Free setting of the weighing capacity, within the display resolution of 1/60,000, isavailable.
 Available weighing units are kg (kilogram), g (gram), Ib (pound), oz (ounce), andpcs (pieces for the counting mode).
 The counting mode easily counts the number of objects of the same weight.
 The comparator mode compares the displayed value (weight value) with thepreviously set comparator values and indicates the results by the large and bright
LED display. The optional comparator relay output (SW-03) can output the results
as a relay signal.
 The auto-tare function, used with the comparator mode, automatically tares anddisplays “OK” for a certain amount of sample and repeats this process for the next
weighing.
 Using the optional RS-232C serial interface (SW-03) or optional RS-422/485 serialinterface (SW-04) provides serial data output of the weight value and relay output of
the comparison result.
 The optional sealed lead acid (SLA) battery (HC-02i) can be installed, allowing thescale to be used where an AC power source is not available.
 SW-11 provides the function of a display stand or wall-mount bracket for the SW-D.

5

4. UNPACKING
Unpack the indicator carefully and keep the packing material if you are likely totransport the indicator again in the future.
When unpacking, check whether all of the following items are included:

Indicator

Model label(1083013259-2)
To be attached. SW-D

CAPACITYSECTION

Write the scale’s ratings, etc..

SW-D
Set up instructions(this manual)

SW series
Instruction manual

SW-D

SW series

SET UP INSTRUCTIONSINSTRUCTION MANUAL

Main power cord
Please confirm that the main
power type is correct for yourlocal voltage and receptacle.

65. INSTALLATION and PRECAUTIONS
5.1. Precautions for Installing the Indicator
DANGER Ground the indicator so that the user will not be subjected an electric shock.
 Do not handle the main power cord with wet hands.
 The AC plug is not water-resistant. Install it in an area where it does not get
wet.
 Even when the optional sealed lead acid (SLA) battery (HC-02i) is used, besure to prevent the AC plug from getting wet.
 Do not install the indicator where there is flammable or corrosive gas present.
 Do not pull, fold or arrange cables forcibly.
 Use shielded cables for all connections.
 Ground the platform to be connected to the indicator to avoid a risk of electric
shock.

Read the section “PRECAUTIONS” in the instruction manual of the SW tounderstand the conditions and precautions for installation, operation and cleaning
the indicator.

5.2. Connecting to the load cell
1. Disconnect the main power cord of the scale from the power source.
2. On the bottom of the display, remove the screw protection caps on the load cellpanel. Loosen the four screws and open the load cell panel.
3. Loosen the cable clamp locking nut on the load cell panel and insert the load cell
cable through the cable clamp.
Accommodates load cell cable diameters of φ3.5 to 7.0 mm.
Load cell
cable
Locking
nut
Cable
clamp
Load cellpanel
74. Solder the wires to the pin contacts of the connector.
 A 5 m or less load cell cable is recommended as the SW-D uses a 4 wire
system (no remote sensing).
SHIELD(EXC-) Negative Excitation(SIG-) Negative Signal In(SIG+) Positive Signal In(EXC+) Positive Excitation
Load cell

Load cell cable

5

SHIELD

4

EXC-

3

SIG-

2

SIG+

1

EXC+

Pin contacts

5. Tighten the cable clamp locking nut to secure the load cell cable.
6. Close the load cell panel by securing the four screws and cover the screws using
the screw protection caps.

5.3. Verifying Load Cell Output and Input Sensitivity

The input sensitivity is 0.2 µV/division or more. When designing a weighing scale, thefollowing calculation formula should be satisfied.
 The input sensitivity requires a change in output voltage, from the weighing device,
to change the display value as a one digit step. In order to achieve a scale withstable performance, select the input sensitivity as large as possible.

Calculating formula:

0.2 ≤E× B×dA×n

A : Rated capacity of load cell
B : Rated output (mV/V)
d : Minimum division
E : Excitation voltage (mV) = 5000 mV for SW-D
n : Number of load cells

8

Example of calculation:

In the case of designing a scale with a capacity of 60 kg and minimum division of 0.005
kg, using one load cell with a rated capacity of 100 kg and rated output of 1 mV/V:

Rated capacity of Load cell:
Rated output:
Minimum division:
Excitation voltage:
Number of load cells:

5.4. Setting Up the Indicator

A = 100 kg
B = 1 mV/V
5000 × 1 × 0.005= 0.25 > 0.2
d = 0.005 kg
100
×1
E = 5000 mV
n=1
Therefore, there is no problem with this design.

Earth terminal After confirming that nothing is touching the key switches on the indicator, connect
the main power cable to a power source that has an earth ground terminal.

5.5. Parameter Settings and Calibration

When a platform scale is initially set up, it is necessary to set some parameters andcalibrate it using a calibration weight.
See “8. SETTING FUNCTIONS OR UNITS FOR USE” to calibrate the scale.

96. DESCRIPTION OF EACH PART
1. Disp

Load cell panel
Main power cord
Earth terminal

3. STABLE

Load cell connector

2. LEDNET indicator

LO

OK

HI

pcs

ZERO indicator

ON/OFF key

SAMPLE key

STABLE

g

NET

kg

ZERO

oz

Weighing
unitslb

ON/OFFSAMPLE

PRINT

UNITS

4. PRINT key UNITS key

ZERO

TARE

ZERO key

TARE key

Symbols
SymbolDescription

STABLE ○Turns on when the weight value is stable.

NET ○Turns on when the NET weight is displayed.
(The tare operation is in progress.)

ZERO ○

Turns on when zero is displayed.
Turns on when the comparator results are displayed.

Weighing units“pcs”, “g”, “kg”, “oz” and “lb” are available.
A selected unit turns on.
10

Key

Description
6.1.
ON/OFF key
Turns the power ON or OFF.
When turned ON, the scale will be automatically set to zero (power-on zero).
To turn the power OFF, press and hold the key.
Note: If pressing the key does not turn the power ON immediately, keeppressing the key.
6.2.
SAMPLE key
In the counting mode (“pcs”), goes to the unit weight storing mode.
Press and hold to go to the comparator value setting mode.
6.3.
PRINT key
Outputs the weight value to the printer.
In the setting mode, this key is used to increase the value of the selected
blinking digit by one.
6.4.
UNITS key
Switches the weighing unit.
In the setting mode, this key is used to shift the blinking digit to the right.
6.5.

ZERO key

Zeroes the scale and sets the display to zero.
6.6.
TARE key
Subtracts the tare (container) weight placed on the weighing pan.
In the setting mode, this key is used to store the setting value and proceed to
the next step.

11

7. CALIBRATION
When calibrating, refer to this set up instructions. (The contents of SW instruction manualare not the same as this set up instructions.)

7.1. Items of the calibration mode The calibration mode has the following three functions.
• Setting of the weighing capacity, minimum division, and decimal point position.
• Gravity acceleration correction• Calibration using a weight How to enter the calibration mode
1. Make sure that the scale is in the weighing mode (displaying “kg”, “g”, “lb”, “oz” or“pcs”).
2. Remove the screw protection caps on the load cell panel on the bottom of thedisplay. Loosen the four screws and then open the load cell panel. The calibration(CAL) switch is located inside.
3. Press the CAL switch. The indicator displays “Cal5et”. (And proceed to “7.2.”,“7.3.”, or “7.4.”)
Calibration (CAL) switch

Load cell panel

12

The chart below shows the relation between each item and the key operation in thecalibration mode.
Weighing mode
Press the (CAL)
switchSetting of the weighing
capacity, minimum division,and decimal point

Gravity acceleration
correction

Calibration using a weight

Setting of the CF function

Press

or

Power OFF
Press the UNITS key to select an item and press the TARE key to execute it.
To exit from the calibration mode, press the ZERO1 key or the CAL switch. The
indicator turns the power off. (Or press the ON/OFF key to turn the power off.)13

7.2. Setting of the weighing capacity / minimum division / decimal point
positionSet the weighing capacity, minimum division, and decimal point position. Be sure to
perform these setting first when connecting to a load cell (base unit). The indicator does
not require setting each time, if nothing is to be changed.
1．Refer to “7.1. Items of the calibration mode”, to enter
the calibration mode. The indicator displays “Cal5et”.
2. Press the TARE key to enter the setting mode of
the weighing capacity, minimum division, and decimal
point position.
3. The indicator displays the currently set minimum
division and calibration unit. (Display example: “d-1kg”) The blinking digit indicates the minimum
division. Press the PRINT key to select 1, 2, or 5.
The unit indicates the calibration unit. Press the
SAMPLE key to select either “kg” or “lb”.
Press the TARE key to store the currently set
minimum division and calibration unit, and proceed to
the next step.

6. P
re

5. P
re

Press

4. After displaying “Cap”, the indicator displays thecurrently set weighing capacity and decimal point
position. (Display example: “0060.00 kg”) Set theweighing capacity and decimal point position using
the following keys:
MODE
PRINT

Shift the blinking digit to the right.
Increases the number (+1) at theblinking digit.

SAMPLE

Shift the decimal point position.

Press theTARE
key to store the currently
displayed weighing capacity and decimal point
position, and proceed to the next step.

7. P
re

 Available setting range of the resolution is 60,000 or
less. The resolution is the value that divides theweighing capacity by the minimum division.
(Decimal point position is ignored.)
5. After displaying “end”, the indicator displays “Cal5et”,
and the setting is complete.
After setting the weighing capacity, minimum division, and decimal point position, be
sure to perform “7.4. Calibration using a weight”.

14

7.3. Gravity acceleration correction
When the scale is first used or has been moved to another location, it should becalibrated using a calibration weight.
But if a calibration weight is not available, the gravity acceleration correction willcompensate the scale. Change the gravity acceleration value stored in the scale to thevalue of the area where the scale will be used. Refer to the gravity acceleration map at
the end of this manual.
Note: Gravity acceleration correction is not required when the scale will becalibrated using a calibration weight at the place where it is to be used.
1. Refer to “7.1. Items of the calibration mode”, to enter thecalibration mode. The indicator displays “Cal5et”.
2. Press the UNITS key to display “g 9.7985” (gravity
acceleration value).
3. Press the TARE key to enter the gravity acceleration
value setting mode.

8. P
re

9. P
re

4. Change the displayed value using the following keys.
UNITS

To shift the blinking digit to the right.

PRINT

To increase the value of the blinking digit
by one.

5. Press the TARE key. The setting value is stored and“end” is displayed. After displaying “g 9.7985” (gravity
acceleration value), the setting is complete.
6. When calibration using a calibration weight is to be
performed, go to step 3 of “7.4. Calibration using aweight”.
To finish the setting procedure, press the ZERO1 key or
the CAL switch. The indicator turns the power off.

15

Press

Press

7.4. Calibration using a weightThis function adjusts the scale (indicator) for accurate weighing.
Calibration must be done when the SW-D is initially connected to a load cell (base unit)or the combination is changed.
Calibration may also be required according the changes in the environment.
 When the scale (indicator) has been moved.
 When the ambient environment has significantly changed.
 For regular calibration.

Note: Prepare a weight, preferably a weight with the same value as the weighing
capacity of the scale to be calibrated. Note that the calibration weight value
can be changed.
1. Turn the power ON and warm up the scale for at least half an hour.
 Change the function setting “poff” or place something on the weighing pan todisable the auto power-off function.
2. Refer to “7.1. Items of the calibration mode”, to enter
the calibration mode. The indicator displays “Cal5et”.
3. Press the

UNITS

key twice and display “Cal”.

4. Press the TARE key. “Cal 0” is displayed. Confirm
that nothing is placed on the weighing pan and wait for the
STABLE indicator to turn on.
5. Press the TARE key. The scale calibrates the zero
point and displays the value of the calibration weight(SPAN calibration).

10. Presstwice

11. Press

 The calibration weight value is equal to the weighing
capacity. (factory setting) If you want to calibrate only the zero point without
calibrating SPAN, press the ZERO1 key. After“end” is displayed, go to step 9.
6. To calibrate with a weight different from theweighing capacity, change the displayed value
using the following keys.
UNITS

To shift the digit that is blinking
to the right.

PRINT

To increase the value of theblinking digit by one.

 Using a weight with the same value as theweighing capacity is recommended. If otherweights are used, use one with a value greaterthan two-thirds of the capacity.

16

Press

Press

Calibration
weightWeighing pan

7. Place the calibration weight with the same value
as displayed on the weighing pan, and wait for
the STABLE indicator to turn on.
8. Press the TARE key. The scale calibrates SPAN and“end” is displayed. After displaying “Cal”, the setting iscomplete.

Press

9. To finish the setting procedure, press the ZERO1 key or
the CAL switch. The indicator turns the power off.
Note: If the scale will be moved to another location, set the gravity acceleration
value for the new location and calibrate the scale according to the procedure
above. Refer to the previous section to set the gravity acceleration value.

17

8. SETTING FUNCTIONS OR UNITS FOR USE
The indicator has CF function settings to specify the indicator performance.
After installing the indicator, the indicator does not require resetting or changing of thisCF function as far as it is used normally.
If you want to maintain only the needed units, set using this function.
The parameters set in the CF function settings and unit setting are maintained even if thepower is turned OFF.

Item

Parameter

8.1. Setting of the CF function

1. Refer to “7.1. Items of the calibration mode”, to enter thecalibration mode. The indicator displays “Cal5et”.
2. Press the MODE

key three times to display “Cf”.

3. Press the TARE

key to display the setting item.

Press
three times

4. Select the setting item and parameter using the following keys:
UNITS
Proceed to next setting item.
PRINT
Increases the number (+1) at the blinking digit.
(Change the parameter) When the currently stored set is changed, the STABLE
indicator “●” turns off.
 Changing the parameter using the PRINT key at CF1,2, 8 or 12 is not available.
5. Press the TARE

key to store the setting. After displaying“end”, the indicator displays “Unit” and proceeds to the unit
setting mode.
 To proceed, without storing the CF function setting, press
the ZERO
6.

key. After displaying “Unit”, proceed to theuse unit setting mode. (Display example: “1110000 pcs”)
When the unit setting is to be performed, go to step 5 of
“8.2. Setting unit for use”.
When the unit setting is not to be performed, press theZERO1 key. After displaying “Cf”, the setting is complete.

 To adjust to the designed scale specification, set otheritems by following the procedure described above.

18

Press

Press

Press

8.2. Setting unit for use

1．Refer to “7.1. Items of the calibration mode”, to enter thecalibration mode. The indicator displays “Cal5et”.
2. Press the MODE

key three times to display “Cf”.

3. Press the TARE

key to display the setting item.

Press
three times

Press

4. To proceed without storing this setting item, press theZERO key.
After displaying “Unit”, proceed to the unit setting mode.
(Display example: “1110000 pcs”)
5. Select the unit to be used, using the following keys.
UNITS
Select the unit (Shift the blinking unit to theright)PRINT
Set whether the selected unit is available (0)or not available (1). (Change the number of theblinking digit to “0” or “1”.) About the unit setting mode
Each of the display digits are assigned to a unit setting.
(The setting at the first and fourth digit is not available.)
Display
assignment

7

6

5

4

3

2

1

Unit

pcs

g

kg

Set
disabled

oz

lb

Set
disabled

Press

Press

-The unit during selection is indicated.
-With each digit, if setting to “0”, its unit is not available, andif setting to “1”, its unit is available.
(Only when set to “1”, is a unit available.)
Even if a unit is stored, availability for use depends on theweighing capacity and resolution.
Example) When setting to “1010000 pcs”, only “pcs”
and ”kg” are available as units.
6. Press theTARE

key to store the setting. After displaying“end”, the indicator displays “Cf”, and the setting is complete.
 If you want to proceed without storing the setting of the unit
setting mode, press the ZEROthe setting is complete.

key. After displaying “Cf”,

19

Press

8.3. CF Function list
Item
Used internally

Parameter

Cf 1

 3

Not available for setting change

Used internally

Cf 2

 0

Not available for setting change

Zero tracking

Cf 3

0

 1
Power on zero
0
range
1
Cf 4
 2
3
Calculation of tare
0
Cf 5
 1
Stable indicator0of during tare
Cf 6
 1
Tare clear by0
zero operation
 1

Cf 7

Used internally

Cf 8

Minus andprinting limit

Cf 9Auto tare function

Cf10

Stable detection
range

Description

Not available for setting of function“trc” (It is fixed to “trc 1”)
Available for setting of function “trc”
None±10% of weighing capacity±50% of weighing capacity±100% of weighing capacity
Tare by displayed division
Tare by internal countDuring tare, stable indicator has nofunction
During tare, stable indicator turns on
Tare is not cleared by zero operation

Not available for setting change 0

Non-limit

1

Limit has function0

No auto tare function
Auto tare function

Cf11

 1

Not available for setting of function“5t-b” (It is fixed to “5t-b 0”)
Available for setting of function “5t-b”

Cf12

 0

Not available for setting change

Used internally0

Calibration zeroreference

Tare is cleared by zero operation

 3

 1

When “Cf 3 0”, “trc”
has no function.

Refer to below note.
When “Cf10 0”, “at”
“at-t” “at-f” has nofunction.
When “Cf11 0”, “5tb” has no function.

 Factory setting

Note: About minus and printing limit setting (“Cf 9”)
When setting “Cf 9” to “1”, the indicator has the following limit.
When the total weight is -19d or less, the indicator displays “-e”.
(*d = minimum division)
When the total weight is minus, the indicator does not perform the printing.

20

9. OPTIONS For details of HC-02i, SW-03, and SW-04, refer to the “OPTIONS” in the SW
series instruction manual.

9.1. SW-11 Display stand / Wall-mount SW-11 provides the function as a display stand or a wall-mount for the SW-D.

Attaching to SW-11
1. Attach the SW-11 to the display using the four screws provided.
2. Adjust the display angle using the two clamps on the SW-11.
After loosening the two clamps on the SW-11, adjust the display angle, and retighten the two clamps.

Clamp

SW-11

3. If you want to use the SW-11 as wall-mount bracket, after fixing only the stand part of
the SW-11 on the flat wall by using the screws, attach the display to the SW-11.
(These screws are not provided.)
For details of the installing dimension, refer to “11.2. External dimensions”.

21

10. MAINTENANCE
10.1. Notes on maintenance Do not disassemble the indicator. Contact your local A&D dealer if the indicatorneeds service or repair.
 Use the original packaging for transportation.
 Do not use organic solvents to clean the indicator. Use a warm lint free cloth
dampened with a mild detergent.
 Calibrate the indicator periodically to maintain the weighing accuracy.

10.2. Error codes
Overload error

Over range error

Unit weight error

CAL error

Low battery

Other

Indicates that an object beyond the weighing capacity
has been placed on the weighing pan.
Remove the object from the weighing pan.

When setting “Cf 9” to “1”, indicates that total weight is–19d or less. (d = minimum division)(In the case of turning on the stable indicator, setting thedisplay to zero by pressing the ZERO key is available.)

Indicates that the sample weight is too light to set the unit
weight in the counting mode.

Indicates that the calibration procedure is canceled
because the calibration weight is too light.
Check that the weighing pan is installed properly and themass of the calibration weight.
Indicates that the HC-02i battery is depleted.
Charge the battery immediately.

There may be an internal malfunction.

(∗ indicates an error number.)Note: If the error persists or other errors occur, contact your local A&D dealer.

2211. SPECIFICATIONS
11.1. Specifications
Input sensitivity
Input signal range
Load cell excitation
voltage
Load cell drive capacity
Temperature ZerocoefficientSpanlinearity
Maximum display
resolution
Display
Display update
Operating temp.
Power supply
Dimension
Weight0.2 µV / d min. (d: minimum division)-16 mV ~ 16 mV
5 V DC ±5% (no remote sensing)
Up to 4 x 350 ohm load cells±(0.2 µV ±0.001% of zero adjustment voltage) / °C (typ.)±0.001% / °C of reading (typ.)±0.01 % of full scale
60,000Weight display: 7 segment LED display (Character height 14.6 mm)
Comparison result: red / yellow / green / yellow / red LED
Approximately 10 times per second-10°C ~ 40°C / 14°F ~ 104°F
AC main (100V +10%/-15%, 50/60 Hz 20VA)
or SLA Battery (option), available use time approximately 70
hours*
Main power cord length: Approximately 3 m
255 (W) x 119 (D) x 179 (H) mm
Approximately 2.1 kg* In the case of one 350 ohm load cell, no other option, and turning on comparator LED(The available use times depend on the condition of the battery, how the battery is used
and the environment.)
Options
HC-02iSLA Sealed Lead Acid battery (YUASA NP4-6 recommended)

SW-03

RS-232C / Relay output

SW-04

RS-422 / 485

SW-11

Display stand / Wall-mountNote: The options, SW-03 and SW-04, can not be used at the same time.

23

11.2. External dimensions
255

118.9

160

47.5

52.5

158

50

52.5

47.5

20.5

4-M4 depth 8mm● When attaching to SW-11
315

135.9

65
50

152.5
197.5
90

283

6-φ5

160

4-M4 x 8 truss screw

110

70
250

Unit: mm24

12. GRAVITY ACCELERATION MAP
Values of gravity at various locations
Amsterdam
AthensAuckland NZ
BangkokBirmingham
Brussels
Buenos Aires
Calcutta
Cape Town
ChicagoCopenhagen
CyprusDjakarta
Frankfurt
Glasgow
HavanaHelsinki
KuwaitLisbonLondon (Greenwich)
Los Angeles
Madrid

9.813 m/s2
9.807 m/s2
9.799 m/s2
9.783 m/s2
9.813 m/s2
9.811 m/s2
9.797 m/s2
9.788 m/s2
9.796 m/s2
9.803 m/s2
9.815 m/s2
9.797 m/s2
9.781 m/s2
9.810 m/s2
9.816 m/s2
9.788 m/s2
9.819 m/s2
9.793 m/s2
9.801 m/s2
9.812 m/s2
9.796 m/s2
9.800 m/s2

ManilaMelbourne
Mexico City
Milan
New York
Oslo
OttawaParis
Rio de Janeiro
Rome
San Francisco
Singapore
Stockholm
SydneyTaichung
TainanTaipeiTokyo
Vancouver, BC
Washington DC
Wellington NZ
Zurich25

9.784 m/s2
9.800 m/s2
9.779 m/s2
9.806 m/s2
9.802 m/s2
9.819 m/s2
9.806 m/s2
9.809 m/s2
9.788 m/s2
9.803 m/s2
9.800 m/s2
9.781 m/s2
9.818 m/s2
9.797 m/s2
9.789 m/s2
9.788 m/s2
9.790 m/s2
9.798 m/s2
9.809 m/s2
9.801 m/s2
9.803 m/s2
9.807 m/s2

World map

26

THIS PAGE INTENTIONALLY LEFT BLANKBack cover: A&D Global

A&D Company, Limited

3-23-14 Higashi-Ikebukuro, Toshima-ku, Tokyo 170-0013, JAPAN
Telephone: [81] (3) 5391-6132
Fax: [81] (3) 5391-1566

A&D ENGINEERING, INC.

47747 Warm Springs Blvd, Fremont, California 94539, U.S.A.
Tel: [1] (800) 726-3364
Weighing Support:[1] (888) 726-5931

A&D INSTRUMENTS LIMITED

Inspection Support:[1] (855) 332-8815

Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire
Telephone: [44] (1235) 550420
Fax: [44] (1235) 550485

OX14 1DY

United Kingdom

A&D AUSTRALASIA PTY LTD

32 Dew Street, Thebarton, South Australia 5031, AUSTRALIA
Telephone: [61] (8) 8301-8100
Fax: [61] (8) 8352-7409

A&D KOREA Limited

한국에이.엔.디(주)OOO A&D RUS

OOO "ЭЙ энд ДИ РУС"서울특별시 영등포구 국제금융로6길33 (여의도동) 맨하탄빌딩 817 우편 번호 07331( 817, Manhattan Bldg., 33. Gukjegeumyung-ro 6-gil, Yeongdeungpo-gu, Seoul, 07331 Korea )전화: [82] (2) 780-4101
팩스: [82] (2) 782-4264

Почтовый адрес:121357, Российская Федерация, г.Москва, ул. Верейская, дом 17Юридический адрес: 117545, Российская Федерация, г. Москва, ул. Дорожная, д.3, корп.6, комн. 8б
( 121357, Russian Federation, Moscow, Vereyskaya Street 17 )тел.: [7] (495) 937-33-44
факс: [7] (495) 937-55-66

A&D Instruments India Private Limited

D-48( D-48, Udyog Vihar, Phase- , Gurgaon - 122016, Haryana, India ): [91] (124) 4715555: [91] (124) 4715599

A&D SCIENTECH TAIWAN LIMITED.

A&D台灣分公司 艾安得股份有限公司

台湾台北市中正區青島東路 5 號 4 樓
( 4F No.5 Ching Tao East Road, Taipei Taiwan R.O.C. )Tel : [886](02) 2322-4722
Fax : [886](02) 2392-1794

A&D INSTRUMENTS (THAILAND) LIMITEDบริษัท เอ แอนด์ ดี อินสทรูเม้นท์ (ไทยแลนด์) จำกัด
168/16 หมู่ที่ 1 ตำบลรังสิต อำเภอธัญบุรี จังหวัดปทุมธานี 12110 ประเทศไทย
( 168/16 Moo 1, Rangsit, Thanyaburi, Pathumthani 12110 Thailand )Tel : [66] 20038911