*To those who wish to double
the production efficiency

ISO9000 CERTIFIED...Clearly a Better Value
http://www.aandd.jp

Learn Exactly How to Improve Your
Production-line Weighing System!
Do you struggle with any of the following problems…?
• Weighing is slow.

• Instruments take up space.

• Weighing sensors do not last long.

• Display units are redundant.

• Cabling is complex and expensive.
If your answer is “yes,” the AD-4212C Series ofelectromagnetic digital load cells (EM-DLC) aredefinitely worth trying. Because of…

Fast Stabilization of 0.5 SecondsBreeze breaki

Using our field-proven, Compact Super Hybrid Sensor (C-SHS)
technology, the AD-4212C realizes a weighing speed of 0.5 seconds
or less for 1 mg resolutionii (1.3 seconds for 0.1 mg resolution).

Upper casepending* iii Patent* For weighing of up to 30 g excluding the tare

Please visit our video library at www.aandd.jp to see a demonstration.

59mm

Compact Size with a 59 mm Width
Again, thanks to the C-SHS, the width of the AD-4212C is just 59mm, making it ideal for installation in narrow spaces.

Super HybridSensorExternalCalibration

RS-232CInterface

Smart Range

Software

IP65

Installation images
30mm
AD-4212C

Weighing panJig

High Durability and Protection
Our tests have shown that the AD-4212C can easily withstand 10 million repeated loadings or more. In
addition, the AD-4212C incorporates our patented shock absorber under the weighing pan. It copeswith movement in all directions so that the sensor is protected from actuator malfunctions.

IP65 Dust and Water-proof Construction

RS-232C cable (removable)

The AD-4212C is well suited for powder or liquid weighing. The
IP65 construction protects it from any accidental spilling inmeasurement.

Direct Connection to a Panel Computer,PC, or PLC
The AD-4212C has a sophisticated analog-to-digital converter
inside. It can therefore output digital data directly to an external
device via its standard RS-232C interface. Whether to add adisplay unit, such as the AD-8923-CC/BCD or AD-8922A (bothsold separately), is completely up to you.

CC-Link Connection through the AD-8923-CC
The optional remote controller, the AD-8923-CC,iii is equipped with a CC-Link interface.iv It can transmit thedata received from the AD-4212C to a PLC using CC-Link, in addition to displaying the weighing results,changing response speed, and performing calibration.
* iii The AD-8923-CC and AD-4212C can share the same power supply.

* iv If you prefer a BCD output, the AD-8923-BCD is also available.

PLC

AD-8923-CC (sold separately)

Other Features
Securable to a base by replacing leveling feet with screws
High-speed data transmission of up to 50 times/second
Easy calibration and response speed adjustment using the provided
WinCT-AD4212C software
Also comes with a free calibration mass (200 g, OIML class E2 equivalent)
LAN communication using the AD-8526 serial/Ethernet converter (sold separately)

Possible applications include the management of filling amounts (e.g. electrolyte
in batteries or medicines in capsules), management of coating amounts (e.g.
liquid resin on LEDs, resist ink on FPDs, grease on bearings, or solder paste onelectronic parts), control of dispensed amounts (e.g. ink from ink-jet printers),
and many more!

Specifications

AD-4212C-301
51 g / 320 g

Weighing capacity

AD-4212C-300

AD-4212C-600

AD-4212C-3100

AD-4212C-3000

AD-4212C-6000

320 g

620 g

510 g / 3200 gv

3200 g6200 gv

Minimum weighing value (d)0.0001 g / 0.001 g

0.001 g0.001 g0.001 g / 0.01 g

0.01 g0.01 gRepeatability (Standard Deviation)

0.0002 g / 0.001 g

0.001 g0.001 g0.002 g / 0.01 g

0.01 g0.01 g±0.002 g±0.002 g±0.005 g±0.02 g±0.02 g±0.04 gd = 0.0001 g : 1.3 sec.vi

0 – 30 g : 0.5 sec.

0 – 30 g : 0.5 sec.

d = 0.001 g : 1.3 sec.vi

0 – 300 g : 0.5 sec.

0 – 300 g : 0.5 sec.

d = 0.001 g : 1.0 sec.

30 – 320 g : 1.0 sec.

30 – 620 g : 1.0 sec.

d = 0.01 g : 1.0 sec.

300 – 3200 g : 1.0 sec.

300 – 6200 g : 1.0 sec.

Linearity
Stabilization time
(when set to FAST under a good environment)

10Display refresh ratevii- 50 times/secondvii

Bi-directional, 2400 - 19200 bps

I/O unit (RS-232C)±2 ppm/˚C (10 ˚C to 30 ˚C / 50 ˚F to 86 ˚F)Sensitivity drift
Operating environment5 ˚C to 40 ˚C (41 ˚F to 104 ˚F), 85%RH or less (no condensation)

Calibration mass provided

200 g (equivalent to OIML Class E2)

Applicable calibration mass value

Weighing unit

50 g, 100 g

50 g, 100 g

50 g, 100 g

50 g, 100 g

50 g, 100 g

200 g vii, 500 g

200 g vii, 300 g

200 g vii, 300 g

200 g vii, 300 g

200 g vii, 300 g

200 g vii, 300 g

1000 g, 2000 g

400 g, 500 g

400 g, 500 g

400 g, 500 g

3000 g, 4000 g

600 g

1000 g, 2000 g

1000 g, 2000 g

5000 g, 6000 g

3000 g3000 gDimensions59(W) X 231(D) X 91(H) mm

Weighing pan

50 X 50 mm

Net weight

Approx. 1.6 kg
Approx. 10 m

Connection cableAC adapter

Power supply

Approx. 11VA (supplied to the AC adapter)

Power consumption

range function: The display will switch to the standard range automatically when the value exceeds 51 g/510 g but return to the precision range by performing RE-ZERO (tare).
* vvi Smartthe precision range is used
* vii When
* Factory setting

Accessories

Dimensions (mm/inches)
Static eliminator
Electrostatic fieldmeterTweezers for calibration mass
Compact printerRemote controller
BCD output interface viii
Remote controller (BCD)
Remote controller (CC-Link)
Serial/Ethernet converter
Serial/USB converter
RS-232C output cable (10 m) ix
RS-232C output cable (5 m) ix
RS-232C output cable (2 m) ix

6.5/0.25

50/1.96
16/0.6248.8/1.92
30/1.18

13/0.5118.8/0.74

59/2.3246±0.2/1.81±0.007

2-ø4.3/0.16
3-M6

4-M4

46/1.81231/9.09
254/9.99

Exclusively for the AD-8922A* viii* ix Exclusively for the AD-4212C

50/1.9614.2/0.55

AD-1683
AD-1684
AD-1689
AD-8121B
AD-8922A
AD-8922A-01
AD-8923-BCD
AD-8923-CC
AD-8526
AX-USB-9P
AX-KO3590-1000
AX-KO3590-500
AX-KO3590-200

97/3.81

1/0.03254/9.99

90.5±1/3.56±0.03

9.3/0.36

99/3.89

AD-8922A

80.5/3.16

Materials
Weighing pan = SUS316
Breeze break = SUS304
Upper case = Zinc die-cast (acrylic coating on epoxy basecoating)AD-1683

AD-1684...Clearly a Better Value
3-23-14 Higashi-lkebukuro, Toshima-ku,Tokyo 170-0013 JAPAN
Telephone:[81](3) 5391-6132 Fax:[81](3) 5391-6148
http://www.aandd.jp
1756 Automation Parkway, San Jose, CA 95131 U.S.A.
Telephone:[1](408) 263-5333 Fax:[1](408) 263-0119

A&D Australasiaia Ptyy Ltd.
d.
32 Dew Street, Thebarton, South Australia 5031 AUSTRALIA
Telephone:[61](8) 8301-8100 Fax:[61](8) 8352-7409

Unit 24/26 Blacklands Way Abingdon Business Park,Abingdon, Oxon OX14 1DY UNITED KINGDOM
Telephone:[44](1235) 550420 Fax:[44](1235) 550485

Vereyskaya str.17, Moscow, 121357 RUSSIA
Telephone: [7] (495) 937-33-44 Fax: [7] (495) 937-55-66<German Sales Office>509 Udyog Vihar Phase V
Gurgaon-122 016, Haryana, INDIA
Telephone: [91](124) 471-5555 Fax: [91](124) 471-5599

Hamburger Straße 30 D-22926 Ahrensburg GERMANY
Telephone:[49](0) 4102 459230 Fax:[49](0) 4102 459231

A&D
D Instruments
Inst uments India Pr
Privatee Limited
mited

Manhattan Bldg. 8F, 36-2 Yoido-dong, Youngdeungpo-gu, Seoul, KOREA
Telephone:[82](2) 780-4101 Fax:[82](2) 782-4280*AD4212C Series-ADCC-01-PR1-11203