 Store the comparator limits into the specified memory number. (No reply for f20-1.)

FS-i Series Check Weighing Scales

Command M L,0 1,+ 0 0 1 2 0 0,

+ 0 0 0 9 0 0, CR LF

ReplyI,0 1,+ 0 0 1 2 0 0,

+ 0 0 0 9 0 0, CR LF

This manual describes the additional functions and corrections to the instruction manual
WM+PD4001332 and 1WMPD4001368. Please note that the new functions are available forthe latest version of the FS-i series only.

Command M L,0 1,+ 0 0 1 2 0 0,

+ 0 0 0 9 0 0, CR LF

1. AUTO-TARE FUNCTION

Reply,0 1,+ 0 0 1 2 0 0,+ 0 0 0 9 0 0, CR LF

HM L

3-2. OP-04 Specifications
The followings are the corrections and the additional information for the optional interface.

Circuit diagram
5 Shield

5 Frame ground FG

6 Frame ground FG

6 Shield

The frame ground and the shield are connected internally and there is no difference
between them to use.

Multi-connection with Print key mode (new function)
This is one of applications when a PC and more than one FS-i are connected. The scale
will prepare the weighing data by pressing the PRINT key first and will send out theprepared data after receiving a command from the PC.
1. Set the Function “f06-5” for all of the scales.
2. Weigh something and wait for the stable annunciator to turn on (the scale having
f18-##).
3. Press the PRINT key, the scale will temporarily store the weight data and turn thePRINT annunciator on.
 The scale does not accept thePRINT

key while the PRINT annunciator on.

4. Send the “@##S” command to the scales from the PC.
5. The scale (f18-##) will respond the command to send the weight data memorized andturn the PRINT annunciator off.
The address ## = 23 (f18-23).
 Request a weight data.

The auto-tare function will be used with comparator function enabled. If the weight value is
within the OK range and stable for seconds of time, then the scale will automatically tare
the weight and show zero. The Function “f22” designates the timing to tare automatically.
 To use the auto-tare function, set the function settings below.
f08-1:
Compare all weighing data (the other setting may be used depending on theapplication).
f21-1:
Auto-tare function enabled.
f22-0~9: Select the timing to tare automatically to avoid the wrong tare operation.
 Take-away check weighing “f24-1”
To use the scale with take-away check weighing (negative comparison), set the Function“f24-1” together with the auto-tare function enabled “f21-1”. The scale operates as“take-away the stuff” Æ “OK and stable” Æ “auto-tare” Æ “take-away the stuff” Æ ······.
In this mode, the comparator results are shown as below.
f07-1:-(|Target|+HI limit) -(|Target|-LO limit)
f07-0:-|Upper limit| -|Lower limit|
Negative value
HIOKNet 0Weight display

LO When the Function “f23-1 Tares the initial (container) weight.” is set:
To start the auto-tare function application, usually the weight of container (filled withstuff) must be tared using the TATE key. When the Function “f23-1 ” is set, the scale
will tare the initial (container weight) weight automatically.
Remove all load on the weighing pan to return to zero point, the tare weight will beautomatically cleared. If the scale could not return to zero point because of the zero shift,press the ZERO key to clear the tare weight.
 If the scale is equipped with the optional RS-232C interface, the OK weighing data only
can be sent out automatically. Set the Function setting “f06-7 Auto-print mode +/- data& OK”.

11-2. Function list

Command @ 2 3 S CR LF

The following function settings are added to the latest FS-i series.
Reply

Reply

@ 2@ 23 S T , + 0 0 1 2 . 3 4 5 _ k g CR LF
3 I CR LF The scale does not have a stored data to send.

 While the PRINT annunciator is on, the scale will send back “@##I” to the commands
other than “@##S” command.

4

Item
Serial interface
Data output
mode

1WMPD4002304

Function number
Description
f06- 0Stream mode
f06- 1
Command mode only f06- 2
Print key mode
f06- 3
Auto-print mode + data
f06- 4
Auto-print mode ± data
1

Optional RS232C/422/485.
Command can beused in all modes.

Item
Function number
Description
Multi-connection with Print key mode The UFC format
Serial interface
f06- 5
is applicable to(RS-422/485)
Data output
f06-2 to 4.
mode
f06- 6
Auto-print mode + data & OK
f06- 7
Auto-print mode +/- data & OK
Zero tracking
f13- 0
Zero tracking OFF
Legal for trademodels cannot
 f13- 1
Zero tracking ON, 0.5d/sec
set f13-2 and 3.
f13- 2
Zero tracking ON, 1.0d/sec
f13- 3
Zero tracking ON, 2.0d/sec
Auto-tare f21- 0
Auto-tare function disabled.
functionf21- 1
Auto-tare function enabled.
Auto-tare timing
f22- 0
Immediately after OK and stable
Timing to tareautomaticallyf22- 1
0.5 sec. after OK and stable
after the comp.
 f22- 2
1.0 sec. after OK and stable
OK and weightf22- 3
1.5 sec. after OK and stablestable.
f22- 42.0 sec. after OK and stable
To be used with
f22- 5
2.5 sec. after OK and stablef21-1.
f22- 6
3.0 sec. after OK and stablef22- 74.0 sec. after OK and stablef22- 85.0 sec. after OK and stablef22- 96.0 sec. after OK and stableAuto-tare for the  f23- 0
Function disabled
Automatic
initial weight
operation.
f23- 1Tares the initial (container) weight.
Normal/Negative  f24- 0
Normal comparison
comparison
f24- 1Negative comparison for take-away Factory setting

2-2. OP-03 Specifications
The followings are the corrections and the additional information for the optional interface.

Circuit diagram
4 Signal ground SG

4 Signal ground SG

5 Shield

5 Frame ground FG

6 Frame ground FG

6 Shield

The frame ground and the shield are connected internally and there is no difference
between them to use.

Example for the weighing unit “lb-oz” (added)
Weighing data “lb-oz” (+)

S T , + 0 0 1 L 0 1 . 6 _ o z CR LF

Out of range “lb-oz” (+)

O L , + 9 9 9 L 9 9 . 9 _ o z CR LF

Data output mode (f06)
The commands can be used in all data output mode (f06-1 is command mode only).
As the new modes, the f06-5, f06-6 and f06-7 are added to the Function “f06”.
 Auto-print Mode + data and OK (f06-6)
Data is sent if the weight display is stable at +5d (d = weighing display division) andabove, and the comparator result OK in addition. The next transmission cannot occur
until after the weight display falls below +5d.
 Auto-print Mode +/- data and OK (f06-7)
Data is sent if the weight display is stable at ±5d (d = weighing display division) andabove/below, and the comparator result OK in addition. The next transmission cannot
occur until after the weight display falls between –5d and +5d.
 Multi-connection with Print key mode (RS-422/485) (f06-5)
This is one of applications when a PC and more than one FS-i (RS-422/485 installed)connected. Refer to “Multi-connection with Print key mode” in detail.

2-3. Command Mode
Examples of command and reply(“_” shows “Space” (20H).)

There are corrections to the instruction manual 1WMPD4001368.
 Set a HI limit value or upper limit weight. (No reply for f20-1.)
When f07-0 or f07-1 is set, the command should have “+” and 6 digit number without
decimal point.
When f07-0 or f07-1 is set, the command should have “+/-” (“+” only for f07-1) and 6digit number without decimal point.
 Set a LO limit value or lower limit weight. (No reply for f20-1.)
When f07-0 or f07-1 is set, the command should have “+” and 6 digit number without
decimal point.
When f07-0 or f07-1 is set, the command should have “+/-” (“+” only for f07-1) and 6digit number without decimal point.
When f07-2 is set, the command should have “+” and 5 digit number without decimal
point.

Data format
 Example of data

CommandHI,+00

100 CR LF Assume 2 decimal place number.

Out of range “kg” (+)

O L, + 9 9 9 9 9 9 9 9 _ k g CR LF

Reply

HI,

+00

100 CR LF 1.00% will be set as LO limit %.

Out of range “kg” (+)

O L, + 9 9 9 9 . 9 9 9 _ k g CR LF

CommandL O,+00100 CR LF Assume 2 decimal place number.

ReplyL O,+00

100 CR LF 1.00% will be set as LO limit %.

The position of decimal point is different according to the model and/or the Function“f02”.
23

Item
Function number
Description
Multi-connection with Print key mode The UFC format
Serial interface
f06- 5
is applicable to(RS-422/485)
Data output
f06-2 to 4.
mode
f06- 6
Auto-print mode + data & OK
f06- 7
Auto-print mode +/- data & OKZero tracking
f13- 0
Zero tracking OFF
Legal for trademodels cannot
 f13- 1
Zero tracking ON, 0.5d/sec
set f13-2 and 3.
f13- 2
Zero tracking ON, 1.0d/sec
f13- 3
Zero tracking ON, 2.0d/sec
Auto-tare f21- 0
Auto-tare function disabled.
functionf21- 1
Auto-tare function enabled.
Auto-tare timing
f22- 0
Immediately after OK and stable
Timing to tareautomaticallyf22- 1
0.5 sec. after OK and stable
after the comp.
 f22- 2
1.0 sec. after OK and stable
OK and weightf22- 3
1.5 sec. after OK and stablestable.
f22- 42.0 sec. after OK and stable
To be used with
f22- 5
2.5 sec. after OK and stablef21-1.
f22- 6
3.0 sec. after OK and stablef22- 74.0 sec. after OK and stablef22- 85.0 sec. after OK and stablef22- 96.0 sec. after OK and stableAuto-tare for the  f23- 0
Function disabled
Automaticinitial weight
operation.
f23- 1Tares the initial (container) weight.
Normal/Negative  f24- 0
Normal comparison
comparison
f24- 1
Negative comparison for take-away Factory setting

2-2. OP-03 Specifications
The followings are the corrections and the additional information for the optional interface.

Circuit diagram
4 Signal ground SG

4 Signal ground SG

5 Shield

5 Frame ground FG

6 Frame ground FG

6 Shield

The frame ground and the shield are connected internally and there is no difference
between them to use.

Example for the weighing unit “lb-oz” (added)
Weighing data “lb-oz” (+)

S T , + 0 0 1 L 0 1 . 6 _ o z CR LF

Out of range “lb-oz” (+)

O L , + 9 9 9 L 9 9 . 9 _ o z CR LF

Data output mode (f06)
The commands can be used in all data output mode (f06-1 is command mode only).
As the new modes, the f06-5, f06-6 and f06-7 are added to the Function “f06”.
 Auto-print Mode + data and OK (f06-6)
Data is sent if the weight display is stable at +5d (d = weighing display division) andabove, and the comparator result OK in addition. The next transmission cannot occur
until after the weight display falls below +5d.
 Auto-print Mode +/- data and OK (f06-7)
Data is sent if the weight display is stable at ±5d (d = weighing display division) andabove/below, and the comparator result OK in addition. The next transmission cannot
occur until after the weight display falls between –5d and +5d.
 Multi-connection with Print key mode (RS-422/485) (f06-5)
This is one of applications when a PC and more than one FS-i (RS-422/485 installed)connected. Refer to “Multi-connection with Print key mode” in detail.

2-3. Command Mode
Examples of command and reply(“_” shows “Space” (20H).)

There are corrections to the instruction manual 1WMPD4001368.
 Set a HI limit value or upper limit weight. (No reply for f20-1.)
When f07-0 or f07-1 is set, the command should have “+” and 6 digit number without
decimal point.
When f07-0 or f07-1 is set, the command should have “+/-” (“+” only for f07-1) and 6
digit number without decimal point.
 Set a LO limit value or lower limit weight. (No reply for f20-1.)
When f07-0 or f07-1 is set, the command should have “+” and 6 digit number without
decimal point.
When f07-0 or f07-1 is set, the command should have “+/-” (“+” only for f07-1) and 6digit number without decimal point.
When f07-2 is set, the command should have “+” and 5 digit number without decimalpoint.

Data format
 Example of data

CommandHI,+00

100 CR LF Assume 2 decimal place number.

Out of range “kg” (+)

O L, + 9 9 9 9 9 9 9 9 _ k g CR LF

ReplyHI,+00

100 CR LF 1.00% will be set as LO limit %.

Out of range “kg” (+)

O L, + 9 9 9 9 . 9 9 9 _ k g CR LF

CommandL O,+00100 CR LF Assume 2 decimal place number.

Reply

L O,

+00100 CR LF 1.00% will be set as LO limit %.

The position of decimal point is different according to the model and/or the Function“f02”.
23

 Store the comparator limits into the specified memory number. (No reply for f20-1.)

FS-i Series Check Weighing Scales

Command M L

,

0 1,

+ 0 0 1 2 0 0,

+ 0 0 0 9 0 0, CR LF

Reply

I,0 1,+ 0 0 1 2 0 0,

+ 0 0 0 9 0 0, CR LF

This manual describes the additional functions and corrections to the instruction manual
WM+PD4001332 and 1WMPD4001368. Please note that the new functions are available forthe latest version of the FS-i series only.

Command M L,0 1,+ 0 0 1 2 0 0,

+ 0 0 0 9 0 0, CR LF

1. AUTO-TARE FUNCTION

Reply

,0 1,

+ 0 0 1 2 0 0,

+ 0 0 0 9 0 0, CR LF

HM L

3-2. OP-04 Specifications
The followings are the corrections and the additional information for the optional interface.

Circuit diagram
5 Shield

5 Frame ground FG

6 Frame ground FG

6 Shield

The frame ground and the shield are connected internally and there is no difference
between them to use.

Multi-connection with Print key mode (new function)
This is one of applications when a PC and more than one FS-i are connected. The scale
will prepare the weighing data by pressing the PRINT key first and will send out theprepared data after receiving a command from the PC.
1. Set the Function “f06-5” for all of the scales.
2. Weigh something and wait for the stable annunciator to turn on (the scale having
f18-##).
3. Press the PRINT key, the scale will temporarily store the weight data and turn thePRINT annunciator on.
 The scale does not accept thePRINT

key while the PRINT annunciator on.

4. Send the “@##S” command to the scales from the PC.
5. The scale (f18-##) will respond the command to send the weight data memorized andturn the PRINT annunciator off.
The address ## = 23 (f18-23).
 Request a weight data.

Reply@ 2@ 2

 To use the auto-tare function, set the function settings below.
f08-1:
Compare all weighing data (the other setting may be used depending on theapplication).
f21-1:
Auto-tare function enabled.
f22-0~9: Select the timing to tare automatically to avoid the wrong tare operation.
 Take-away check weighing “f24-1”
To use the scale with take-away check weighing (negative comparison), set the Function“f24-1” together with the auto-tare function enabled “f21-1”. The scale operates as“take-away the stuff” Æ “OK and stable” Æ “auto-tare” Æ “take-away the stuff” Æ ······.
In this mode, the comparator results are shown as below.
f07-1:-(|Target|+HI limit) -(|Target|-LO limit)
f07-0:-|Upper limit| -|Lower limit|
Negative value
HIOKNet 0Weight display

LO When the Function “f23-1 Tares the initial (container) weight.” is set:
To start the auto-tare function application, usually the weight of container (filled withstuff) must be tared using the TATE key. When the Function “f23-1 ” is set, the scale
will tare the initial (container weight) weight automatically.
Remove all load on the weighing pan to return to zero point, the tare weight will beautomatically cleared. If the scale could not return to zero point because of the zero shift,press the ZERO key to clear the tare weight.
 If the scale is equipped with the optional RS-232C interface, the OK weighing data only
can be sent out automatically. Set the Function setting “f06-7 Auto-print mode +/- data& OK”.

11-2. Function list

Command @ 2 3 S CR LF
Reply

The auto-tare function will be used with comparator function enabled. If the weight value iswithin the OK range and stable for seconds of time, then the scale will automatically tarethe weight and show zero. The Function “f22” designates the timing to tare automatically.

3 S T , + 0 0 1 2 . 3 4 5 _ k g CR LF
3 I CR LF The scale does not have a stored data to send.

 While the PRINT annunciator is on, the scale will send back “@##I” to the commands
other than “@##S” command.

4The following function settings are added to the latest FS-i series.
Item
Serial interface
Data output
mode

1WMPD4002304

Function numberDescription
f06- 0Stream mode
f06- 1
Command mode only f06- 2Print key mode
f06- 3
Auto-print mode + data
f06- 4
Auto-print mode ± data1

Optional RS232C/422/485.
Command can beused in all modes.