WMP-153
Medical Panel PC

USER’S MANUAL

P/N: 205G000WMP1530, V1.0

Copyright© ,2015. All rights reserved
All other brand names are registered trademarks of their respective owners.
The information contained in this document is subject to change without noticeVersion Change History
Date
2015/6/25

ii

Version Description
1.0First release

WMP-153 User’s manual

RemarkJames

Acknowledgments
 Intel® Atom processor is registered trademarks of Intel®
Corporation.
 IBM, PC/AT, PS/2 are trademarks of International Business
Machines Corporation.
®

 Microsoft Windows is a registered trademark of Microsoft
Corporation.

®

 RTL is a trademark of Realtek Semi-Conductor Co., Ltd.
 C&T is a trademark of Chips and Technologies, Inc.
 UMC is a trademark of United Microelectronics Corporation.
 ITE is a trademark of Integrated Technology Express, Inc.
 Intel is a trademark or registered trademark of Intel
Corporation.
 Microsoft Windows is a registered trademark of Microsoft
Corporation.
 Winbond is a registered trademark of Winbond Electronics
Corporation.
All other product names or trademarks are properties of
their respective owners.

WMP-153 User’s manualiii

FCC Class B
This equipment has been tested and found to comply with the limits for aClass B digital device, pursuant to Part 15 of the FCC Rules.
These limits are designed to provide reasonable protection against harmfulinterference when the equipment is operated in a residential environment.
This equipment generates, uses and can radiate radio frequency energy. If
not installed and used in accordance with this user manual, it may cause
harmful interference to radio communications.
Note that even when this equipment is installed and used in accordancewith this user manual, there is still no guarantee that interference will notoccur. If this equipment is believed to be causing harmful interference toradio or television reception, this can be determined by turning theequipment on and off. If interference is occurring, the user is encouraged
to try to correct the interference by one or more of the following measures:Reorient or relocate the receiving antennaIncrease the separation between the equipment and the receiverConnect the equipment to a power outlet on a circuit different fromthat to which the receiver is connectedConsult the dealer or an experienced radio/TV technician for help
Warning:
Any changes or modifications made to the equipment which are
not expressly approved by the relevant standards authority could
void your authority to operate the equipment.
To avoid risk of electric shock, this equipment must only beconnected to a supply mains with protective earth.
Do not modify this equipment without authorization of the
manufacturer.

iv

WMP-153 User’s manualSafety Instructions
Intended use
The WMP-153 is intended to serve as a medical monitor which isdesigned for general purpose for hospital environment and for
diagnosis. It could be used for Surgical, Radiology, PACS (Picture
Archiving Communication Systems), LIS (Lab Information
Systems) and Electronic Medical Record purpose. It shall not be
used for life-supporting system.
WARNING: Critical diagnostic decision must not be based solely
on images displayed by this device
Greeting & Setup
Thank you for purchasing the WMP-153 unit. We wish that thisunit will be durable and reliable in providing your medical
application needs. Please follow the instructions below to ensure
the unit continues to have high performance.
UnpackingAfter opening the carton, there will be a medical panel PC unit
with an accessory box. Examine the contents to see if there aredamages to the unit and if all accessories are present.
Setting up
Please read this manual carefully and remember to keep thismanual for future reference.
Safety Instructions & Cleaning
The unit has undergone various tests in order to comply withsafety standards. Inappropriate use of the open frame unit may
be dangerous. Please remember to follow the instructions below
to insure your safety during the installation and operatingprocess.
Transporting & Placement of unit
1.

When moving the unit on a cart; be very cautious. Quickstops, excessive forces and uneven surfaces may cause
the cart to overturn thus risking the unit to fall to theground.

WMP-153 User’s manualv

2.

If the medical panel PC unit does fall to the ground,
immediately turn the power off and disconnect cords. Thencontact a service technician for repairs. Continual use of
the unit may result cause a fire or electric shock. Also, do
not repair the unit on your own.

3.

Having two or more people transporting the display unit isrecommended. In addition, when installing the unit bysuspending it also requires two or more people.

4.

Before suspending the unit, make sure the material used
for suspension is sturdy and stable. If not properly
suspended, the display unit may fall and cause serious
injury to people standing nearby as well as to the unit
itself.

5.

If you wish to mount the display unit, remember to use
only the mounting hardware recommended by the
manufacturer.

Electrical and Power Source Relatedvi

1.

This medical panel PC unit must operate on a power source
as shown on the specification label. If you are not surewhat type of power supply used in the area, consult yourdealer or local power supplier.

2.

The power cords must not be damaged. Applied pressure,added heat, and tugging may damage the power cord.

3.

The power cord must be routed properly when setup takes
place. We advise that this aspect measure is to prevent
people from stepping on the cords or while the unit issuspended to prevent flying objects from getting tangledwith the unit.

4.

Do not overload the AC outlets or extension cords.
Electrical shocks or fires may occur from overloading.

5.

Do not touch the power source during a thunderstorm.

6.

If your hands are wet, do not touch the plug.

7.

Use your thumb and index finger, grip firmly on the power
cord to disconnect from the electrical socket. By pulling thepower cord, may result in damaging it.

WMP-153 User’s manual

8.

If the unit is not going to be in use for an extended periodof time, remember to disconnect the unit.

9.

The medical panel PC unit uses voltage between
100-240VAC. Connect the unit to a power source with thesame numerical value as shown. Please use only the power
cord provided by the dealer to ensure safety and EMC
compliance.

Various Factors of Environment
1.

Do not insert objects into the openings.

2.

Do not have liquids seep into the internal areas of the
medical panel PC unit.

3.

Having liquids seep in or inserting objects into the unit
may result in electric shocks from taking and/or shortcircuiting the internal parts.

4.

Do not place the medical panel PC unit in the presence ofhigh moisture areas.

5.

Do not install the medical panel PC unit in a wet
environment.

6.

Do not place near unit near heat generating sources.

7.

Do not place the unit in a location where it will come incontact with fumes or steam.

8.

Remember to keep the medical panel PC unit away fromthe presence of dust.

9.

If water has flow in or seep in, immediately disconnect the
open frame unit. Then contact a service technician forrepairs.

Ventilation Spacing
1.

Do not cover or block the openings on the top and backsides of the display unit. Inadequate ventilation may cause
overheating thus reducing the lifespan of the unit.

2.

Unless proper ventilation is present, do not place unit in an
enclosed area; such as a built-in shelf. Keep a minimum
distance of 10 cm between the display unit and wall.

WMP-153 User’s manualvii

Cleaning the unit
1.

Remember to turn off the power source and to unplug thecord from the outlet before cleaning the unit.

2.

Carefully dismount the unit or bring the unit down fromsuspension to clean.

3.

Please use a dry soft cloth to clean the unit.

4.

Take a dry cloth and wipe the unit dry. Remember to avoid
having liquids seep into the internal components and areasof the medical panel PC unit.

Servicing, Repairing, Maintenance & Safety Checks
1.

If the unit is not functioning properly, observe theperformance level of the display closely to determine what
type of servicing is needed.

2.

Do not attempt to repair the medical panel PC unit on yourown. Disassembling the cover exposes users’ to highvoltages and other dangerous conditions. Notify andrequest a qualified service technician for servicing the unit.

3.

To avoid risk of electric shock, this equipment must only beconnected to a supply mains with protective earth.

4.

If any of the following situations occur turn the powersource off and unplug the unit. Then contact a qualified
service technician.
(a) A liquid was spilled on the unit or objects have fallen
into the unit.
(b) The unit is soaked with liquids.
(c) The unit is dropped or damaged.
(d) If smoke or strange odor is flowing out of the operating
unit.
(e) If the power cord or plug is damaged.
(f) When the functions of the unit are dysfunctional.

5.

viii

When replacement parts are needed for the medical panel
PC unit, make sure service technicians use replacement
parts specified by the manufacturer, or those with thesame characteristics and performance as the original
WMP-153 User’s manualparts. If unauthorized parts are used it may result instarting a fire, electrical shock and/or other dangers.
ISO 7000-0434 : Caution, consult ACCOMPANYING
DOCUMENTS.
ISO 7000-1641 : Follow operating instructions orConsult instructions for use.
IEC 60417 -5009 : STAND-BY.
IEC 60417-5031 : Direct current.
EU-wide legislation, as implemented in each
Member State, requires that waste electrical andelectronic products carrying the mark (left) must be
disposed of separately from normal household
waste. This includes monitors and electricalaccessories, such as signal cables or power cords.
When you need to dispose of your display products,
please follow the guidance of your local authority, or
ask the shop where you purchased the product, or if
applicable, follow any agreements made between
yourself.
The mark on electrical and electronic products only
applies to the current European Union Member
States.

WMP-153 User’s manualixWhen networking with electrical devices, the operator is
responsible for ensuring that the resulting system meets therequirements set forth by the following standards:– EN 60601-1 (IEC 60601-1)
Medical electrical equipment
Part 1: General requirements for safety– EN 60601-1-1 (IEC 60601-1-1)
Medical electrical equipment
Part 1-1: General requirements for safety
Collateral standard: Safety requirements for
Medical electrical systems– EN 60601-1-2 (IEC 60601-1-2)
Medical electrical equipment
Part 1-2: General requirements for safety
Collateral standard: Electromagnetic compatibility;
Requirements and tests

Accessory equipment
connected to theanalog and digitalinterfaces must be incompliance with therespective nationallyharmonized IEC
standards (i.e. IEC 60950 for data processing equipment, IEC
60065 for video equipment, IEC 61010-1 for laboratory
equipment, and IEC 60601-1 for medical equipment.)
Furthermore all configurations shall comply with the systemstandard IEC 60601-1-1. Everybody who connects additional
equipment to the signal input part or signal output partconfigures a medical system, and is therefore, responsible thatthe system complies with the requirements of the systemstandard IEC 60601-1-1. The unit is for exclusive interconnection
with IEC 60601-1 certified equipment in the patient environment
and IEC 60XXX certified equipment outside of the patient
environment. If in doubt, consult the technical services
department or your local representative.
x

WMP-153 User’s manualCaution:
DO NOT LEAVE THIS EQUIPMENT IN AN UNCONTROLLEDENVIRONMENT WHERE THE STORAGE TEMPERATURE IS BELOW-20° C (-4° F) OR ABOVE 60° C (140° F). THIS MAY DAMAGE THE
EQUIPMENT.
This equipment shall not be used in life support systems.
The user is not to touch SIP/SOPs and the patient at the sametime.
The patient is the intended operator.
Caution – Use suitable mounting apparatus to avoid risk of injury.
The sound pressure level at the operator’s position according to
IEC 704-1:1982 is no more than 70dB (A).
A) Grounding reliability can only be achieved when the equipment
is connected to an equivalent receptacle marked “Hospital
Only” or “Hospital Grade”.
B) Use a power cord that matches the voltage of the power outlet,
which has been approved and complies with the safety
standard of your particular country.
C) Caution: This adapter Sinpro HPU101-105 is a forming part ofthe medical device
Contact information:
3F, No.14, Prosperity Road II, Science-Based Industrial Park,
Hsinchu, Taiwan 300, R.O.C
TEL: (886) 3 5780000
E-Mail: Sales@wincomm.com.tw

WMP-153 User’s manualxiTable of Contents
Acknowledgments ...................................... iii
FCC Class B ............................................... iv
Safety Instructions ...................................... v

Introduction ............................................1
Product Description .....................................1
Package list ................................................2
Features ....................................................3
Specifications .............................................4
Guidance and manufacturer’s declaration –electromagnetic emissions ...........................6
Guidance and manufacturer’s declaration –electromagnetic immunity ............................7
Guidance and manufacturer’s declaration –electromagnetic immunity ............................9
Immunity ...................................................9
Cleaning/Disinfecting ................................. 11

Getting Started ......................................12
System Set Up.......................................... 12
Dimension ................................................ 13
System View ............................................ 14
Disconnect Device ..................................... 17

BIOS Setup ............................................18
Appendix................................................30
A. Jumper settings and Connectors ............. 30
B. L type Stand (optional kit) ...................... 32
C. Battery Pack Specifications ..................... 33
D. How to disable battery when system hang up-WMP-153 ................................................ 34
E. Scrap Computer Recycling ...................... 35

xii

WMP-153 User’s manualIntroduction
Product Description
The WMP-153 Medical Panel PC is based on Intel ATOM processor,
it accommodates one 2.5” SATA II hard disk drive and up to 8GB
DDR3L SODIMM.
The high brightness LCD, Low noise Fanless solution, integrated
multimedia functions and extensive expansion options makethem the perfect platform upon which to build comprehensive
lifestyle computing applications.
The WMP-153 includes all the features of a powerful computer
into a slim and attractive chassis. The WMP-153 has a 15.6” highbrightness TFT display with 1366 x 768 resolutions.
The WMP-153 is compact, Giga LAN and selectable WLAN network
compatible PC with full safety and medical approval and features
to control a dedicated system with a wide variety of applications.
Combining the WMP-153 into your system can achieve bothcost-saving and efficient improvements.
Common applications include Surgical, Radiology, PACS (Picture
Archiving Communication Systems), LIS (Lab Information
Systems) and Electronic Medical Record. The WMP-153 aredefinitely your perfect choices.

WMP-153 User’s manual1Package list
Before you begin installing your Medical Station, please makesure that the following items have been shipped: The WMP-153 Medical Panel PC unit One DVD containing user's manual, chipset drivers
 Power Adapter x 1 (Mf:Sinpro, type/model: HPU101-105)
 Power cord – Hospital grade used(US type), or other type in
UK, EU…etc.
 Screw x 8 (VESA 75/100 use)2

WMP-153 User’s manualFeatures

Seamless on the front sideEasy wipe surfaces with no internal corners



IP65 at front side and IPX1 for whole systemIntel® Celeron® J1900 2.0 GHz quad-core processor 10W



Projective Capacitive Multi-touch screen (10 fingers)

Two mini PCI-e expansion (One full and one half)

HDD Anti-vibration mechanismFanless solution

Anti-bacteria (MRSA) plastic housing

Built-in Battery backup function (option)Support PCI-E x 1 (option)Support isolated module by different configuration (optional
order configuration)

WMP-153 User’s manual3Specifications[

Hardware Specifications
Display
15.6” 400 nits TFT LCDIntel® Celeron® J1900 2.0 GHz quad-core
CPU Support
processor 10W
Disk Drive Space2.5” Hard Disk Drive (SATA II)
Two Mini PCIe slot (+3.3V loaded 0.5A, total loaded= 1A);
Expansion
One PCI-E x 1 expansion slot (+3.3 loaded 1A,+12V loaded 0.56A, total loaded = 10.02W)
Power Button // Mute // Audio adjustment (+)(-) //brightness (+)(-) // LCD on/off // Clean me(auto
Buttonrelease after 1 minute) // Dis-webcam // Fn

I/O

LCD Specifications
Model Name
Display TypeMax. Resolution
Contrast Ratio
Pixel Pitch (um)Luminance (cd/m2)
Viewing Angle
OperatingTemperature
Brightness Control

4Standard version
1 RS-232 port + 1 RS-232/422/485 port
1 USB 3.0 port + 3 USB 2.0 ports
1 DC-in w/ lock function
1 Gigabit LAN RJ-45 Connectors
1 DVI-I output
Sound:
1 x Line-in
1 x line-out
2 x 2W Speakers on back side

G156XW01 V1
LED backlight LCD
1366 x 768
500 : 1 (Typ)
252 (per one triad) × 252
400 (TYP)
170°(H)160°(V)
0°C~ 40°C (32°F~104°F)
Yes

WMP-153 User’s manualPower Adapter Specifications
Power
Close-frame
MFR
SinproType
HPU101-105
Input Voltage
AC 100 ~ 240 V, 1.25 – 0.5A @ 47 ~ 63 Hz
Output Voltage
DC 12V @ 8.33 A
MTBF
100,000 hrs operation at 25°C
Mechanical Specifications
Architecture
Close-frame
Front Bezel
PET bezel with PCT touch screen
Color
Medical-white
Mounting / Holder
VESA 75/100mm
Construction
3mm ABS + PC TYPE Plastic housing
Dimension (WxHxD)
430 (H) x 300 (V) x 83 (T)Net Weight
5.0 kg (w/o power adapter)
Packing Filler
PE
Environmental Specifications
Operating: 0°C to 40°C (32°F ~104°F)
Temperature
Storage, Transportation: -20°C to 60°C (-4°F~140°F)
Operating:
10% to 90%@ 40°C, non-condensing
Humidity
Storage, Transportation:
10% to 90%, non-condensing
Operating: 15g/0.53 oz, 11 ms, half sine wave
Vibration
Non-operating: 50g/1.76 oz, 11 ms, half sine
wave
Operating: 5 ~ 17 Hz ,Amplitude：0.117 ~ 500Hz ,ShockAcceleration：1.0G
Non-operating:10~55Hz/0.15g, 55~500Hz/2.0g
Operational: up to 3000 m (9842 feet)
Altitudes
Shipping: up to 12192 m (40000 feet)
700 – 1060 hPa (Operation)
Pressure
186 – 1060 hPa (Storage)
186 – 1060 hPa (Transportation)

WMP-153 User’s manual5Input Power Rating
Power Consumption
Touch Screen
TypeInterface
Resolution
Light Transmission
Life TimeFor Adaptor:
AC100~240 V, 1.25 – 0.5A, @47 ~ 63 Hz.
For Unit:
DC 12V, 8.33A
Typical: 54W

Full flat projective capacitive touch panel
Controller with USB interface, 5V100ppi to 25ppi Based WIN7 definition ppi(Pixel per inch)
90% ± 3%
100M times

Guidance and manufacturer’s declaration – electromagnetic
emissions
The model WMP-153 is intended for use in the electromagnetic
environment specified below. The customer or the user of the model
WMP-153 should assure that it is used in such an environment.
Emissions test ComplianceElectromagnetic environment –guidance
RF emissions
The model WMP-153 uses RF
energy only for its internal function.
CISPR 11
Therefore, its RF emissions are very
low and are not likely to cause any
interference in nearby electronicequipment.
RF emissions
The model WMP-153 is suitable foruse in all establishments, including
CISPR 11domestic establishments and those
Harmonic
directly connected to the public
emissionslow-voltage power supply network
IEC 61000-3-2that supplies buildings used for
Voltagedomestic purposes.
fluctuations/
flicker emissions
IEC 61000-3-3
Recommended separation distances between
portable and mobile RF communications equipment and the model
WMP-153
6

WMP-153 User’s manual

The model WMP-153 is intended for use in an electromagnetic
environment in which radiated RF disturbances are controlled. The
customer or the user of the model WMP-153 can help prevent
electromagnetic interference by maintaining a minimum distance
between portable and mobile RF communications equipment(transmitters) and the model WMP-153 as recommended below,
according to the maximum output power of the communications
equipment.
Rated maximum
Separation distance according to
output power of
frequency of transmittertransmittermW
150 kHz to
80 MHz to
800 MHz to
80 MHz800 MHz2,5 GHzd = 1,2
d = 1,2d = 2,3
0,01
0,12
0,12
0,23
0,1
0,38
0,38
0,73
1
1,2
1,2
2,3
10
3,8
3,8
7,3
100
12
12
23
For transmitters rated at a maximum output power not listed above, therecommended separation distanced in meters (m) can be estimatedusing the equation applicable to the frequency of the transmitter, where P
is the maximum output power rating of the transmitter in watts (W)according to the transmitter manufacturer.
NOTE 1 At 80 MHz and 800 MHz, the separation distance for the higher
frequency range applies.
NOTE 2 These guidelines may not apply in all situations. Electromagnetic
propagation is affected by absorption and reflection from structures,
objects and people.
Guidance and manufacturer’s declaration – electromagnetic
immunity
The model WMP-153 is intended for use in the electromagnetic
environment specified below. The customer or the user of the model
WMP-153 should assure that it is used in such an environment.
Immunity test
IEC 60601
ComplianceElectromagnetic
level
environment –test levelguidance

WMP-153 User’s manual7Electrostatic
discharge(ESD)
IEC 61000-4-2

6 kV contact

6 kV contact

8 kV air

8 kV airElectrical fasttransient/burst

2 kV forpower
supply lines

2 kV forpower
supply lines

1 kV for
input/outputlines

1 kV for
input/outputlines

1 kV line(s)to line(s)1 kV line(s)to line(s)2 kV line(s)to earth<5 % UT(>95 % dip in
UT)
for 0,5 cycle

2 kV line(s)to earth<5 % UT(>95 % dip in
UT)
for 0,5 cycle

40 % UT(60 % dip in
UT)
for 5 cycles

40 % UT(60 % dip in
UT)
for 5 cycles

70 % UT(30 % dip in
UT)
for 25 cycles

70 % UT(30 % dip in
UT)
for 25 cycles<5 % UT(>95 % dip in
UT)
for 5 sec

<5 % UT(>95 % dip in
UT)
for 5 sec

IEC 61000-4-4

Surge
IEC 61000-4-5

interruptions
andvoltagevariations
on power
supplyinput lines
IEC
61000-4-11

8

WMP-153 User’s manual

Floors should be
wood, concrete or
ceramic tile. If floors
are covered withsynthetic material, therelative humidity
should be at least 30%.
Mains power quality
should be that of atypical commercial or
hospital environment.

Mains power quality
should be that of atypical commercial or
hospital environment.
Mains power quality
should be that of atypical commercial or
hospital environment.
If the user of themodel WMP-153
requires continued
operation during
power
mains interruptions, it
is recommended thatthe model WMP-153
be powered from anuninterruptible power
supply or a battery.

Powerfrequency(50/60 Hz)
magnetic field

3 A/m

3 A/m

Power frequency
magnetic fields
should be at levels
characteristic of atypical location in a
typical commercial or
IEC 61000-4-8
hospital environment.
NOTE UT is the a.c. mains voltage prior to application of the test level.
Guidance and manufacturer’s declaration – electromagnetic immunity
The model WMP-153 is intended for use in the electromagnetic environment specified below. The
customer or the user of the model WMP-153 should assure that it is used in such an environment.
Immunity
IEC
ComplianceElectromagnetic environment – guidance
60601
level
test level
Portable and mobile RF communications
equipment should be used no closer to any partof the model WMP-153, including cables, than
the recommended separation distancecalculated from the equation applicable to thefrequency of the transmitter.
Recommended separation distance
Conducted RF
IEC 61000-4-6

Radiated RF
IEC 61000-4-3

3 Vrms150 kHz
to 80
MHz

3 V/m
80 MHzto2,5GHzd = 1,2
Vrms

V/md = 1,280 MHz to 800 MHz

d = 2,3

800 MHz to 2,5 GHz

where P is the maximum output power rating of
the transmitter in watts (W) according to thetransmitter manufacturer and d is therecommended separation distance in meters(m).
Field strengths from fixed RF transmitters, as
determined by an electromagnetic site survey, ashould be less than the compliance level ineach frequency range. b
Interference may occur in the vicinity ofequipment marked with the following symbol:

NOTE 1 At 80 MHz and 800 MHz, the higher frequency range applies.
NOTE 2 These guidelines may not apply in all situations. Electromagnetic propagation is affected byabsorption
and reflection from structures, objects and people.

WMP-153 User’s manual9aField strengths from fixed transmitters, such as base stations for radio (cellular/cordless) telephones
and land mobile radios, amateur radio, AM and FM radio broadcast and TV broadcast cannot be
predicted theoretically with accuracy. To assess the electromagnetic environment due to fixed RF
transmitters, an electromagnetic site survey should be considered. If the measured field strength in
the location in which the model WMP-153 is used exceeds the applicable RF compliance level above,the model WMP-153 should be observed to verify normal operation. If abnormal performance isobserved, additional measures may be necessary, such as reorienting or relocating the model
WMP-153.
bOver the frequency range 150 kHz to 80 MHz, field strengths should be less than 3 V/m.

10

WMP-153 User’s manualCleaning/Disinfecting
Steps:
1. Wipe the WMP-153 with a dry clean cloth.
2. Prepare agent per manufacturer’s instructions or hospital
protocol.



Cautions:

Do not immerse or rinse the WMP-153 and its peripherals. If
you accidentally spill liquid on the device, disconnect theunit from the power source. Contact your Biomed regarding
the continued safety of the unit before placing it back inoperation.



Do not spray cleaning agent on the chassis.



Do not use disinfectants that contain phenol.



Do not autoclave or clean the WMP-153 or its peripherals
with strong aromatic, chlorinated, ketone, ether, or Esther
solvents, sharp tools or abrasives. Never immerse electricalconnectors in water or other liquids.

WMP-153 User’s manual11Getting Started
System Set Up
The following is a summary of the steps in setting up the system
for use.
(1). You can fix the system to a mounting fixture using the screw
holes on the sides of the system.
(2). Make any required external connections such as the display,
keyboard, and LAN.
(3). Plug the appropriate end of the power cord into the power
connector on the rear of the system and the plug to anelectrical outlet.
(4). Waiting for 3 seconds then press the power switch on thefront panel of the system once to turn on the system power.
(5). If necessary, run the BIOS SETUP programs to configure thesystem.

Caution:In order to boot up system from USB-CD/DVD drive,please connect USB-CD/DVD drive, turn on computer
power, keep on pressing “F11” key, go into BIOS quickboot menu, select “USB-CD ROM”, WAIT FOR 20
SECONDS, then press enter, system OS will boot up
from USB-CD/DVD drive directly.

12

WMP-153 User’s manualDimension
WMP-153(VESA Mount Screw type: M4)

WMP-153 User’s manual13System View

Front View

Webcam (optional)

Hotkey and LED definition at front panel
Located on Touch screen bottom side, from left to right, front view
1-1. Up. HDD: Yellow
1-2 Down. Battery:
LED 1
LED 2
(Charge/Discharge/Low)(Batt present/ not present)

In charge

blinking

ON

Discharge

OFF

ON

Full charge

ONON

Low battery

blinking

blinking

No battery

OFF

OFF

2. Power (with LED status indicator: ON: Green, OFF: dark)
14

WMP-153 User’s manual

3. Mute
4. Volume adjustment (-)
5. Volume adjustment (+)
6. Brightness (-)
7. Brightness (+)
8. LCD on/off (with LED status indicator: LCD ON: dark, LCD OFF:
Yellow)
9. Clean me (with LED status indicator: ON: Yellow, OFF: dark)a. Keep on contacting 5 seconds to active
b. keep contacting 5 seconds to releasec. auto release after 60 seconds
10. Dis-Webcam: Webcam hotkey disable function
11. Fn: Function key (with LED status indicator: ON: Green, OFF:dark)

I/O parts (WMP-153)

WMP-153 User’s manual

15

16

WMP-153 User’s manualDisconnect Device 

Unplug the power cord from the power
adapter jack to disconnect the device.

WMP-153 User’s manual17

BIOS Setup
BIOS Introduction

The AMI BIOS (Basic Input / Output System) installed in yourcomputer system’s ROM supports Intel processors. The BIOS
provides critical low-level support for a standard device such as
disk drives, serial ports and parallel ports. It also adds virus andpassword protection as well as special support for detailed
fine-tuning of the chipset controlling the entire system.

BIOS Setup
The AMI BIOS provides a Setup utility program for specifying thesystem configurations and settings. The BIOS ROM of the systemstores the Setup utility. When you turn on the computer, the AMIBIOS is immediately activated. Pressing the <Del> key
immediately allows you to enter the Setup utility. If you are alittle bit late pressing the <Del> key, POST (Power On Self Test)
will continue with its test routines, thus preventing you frominvoking the Setup. If you still wish to enter Setup, restart thesystem by pressing the ”Reset” button or simultaneously pressing
the <Ctrl>, <Alt> and <Delete> keys. You can also restart by
turning the system Off and back On again. The following message
will appear on the screen:
Press <DEL> to Enter Setup
In general, you press the arrow keys to highlight items, <Enter>to select, the <PgUp> and <PgDn> keys to change entries, <F1>
for help and <Esc> to quit.
When you enter the Setup utility, the Main Menu screen willappear on the screen. The Main Menu allows you to select fromvarious setup functions and exit choices.

18

WMP-153 User’s manual

MainThis section provides information on the BIOS information, and Battery information

System Date
Set the system date. Use the <Tab> key to switch between data
elements.
System Time
Set the system time. Use the <Tab> key to switch between timeelements.

WMP-153 User’s manual

19Advanced

Trusted Computing
Configuration
Enables or disables BIOS support for security device. O.S. will
not show Security Device. TCG EFI protocol and INT1A interface
will not be available.
ACPI Settings
Enable Hibernation
Enables or Disables System ability to Hibernate (0S/S4Sleep State). This option may be not effective with some OS.
ACPI Sleep State
Select the highest ACPI sleep state the system will enter
when the SISPEND button is pressed.
F81216 Super IO Configuration
Serial Port 1 Configuration
Serial Port
Enable or Disable Serial Port (COM).
Function
20

WMP-153 User’s manual

RS232,RS422,RS485
Change Settings
Select an optimal settings for super IO Device.
Change Settings
Select an optimal settings for Super IO Device.
Set Parameters of Serial Port 1.
Serial Port 2 Configuration
Serial Port
Enable or Disable Serial Port (COM).
Change Settings
Select an optimal settings for super IO Device.
Serial Port 3 Configuration
Serial Port
Enable or Disable Serial Port (COM).
Change Settings
Select an optimal settings for super IO Device.
Serial Port 4 Configuration
Serial Port
Enable or Disable Serial Port (COM).
Change Settings
Select an optimal settings for super IO Device.
Intel(R) Smart Connect Technology
ISCT Support
Enable/Disable Intel Smart Connect
S5 RTC Wake Settings
Wake system From S5
Enable or disable System wake on alarm event. SelectFixedTime, System will wake on the hr::min::sec specified.
Select DynamicTime , System will wake on the current time
+ Increase minute(s).
CPU Configuration
Socket 0 CPU Information
Socket specific CPU Information.
WMP-153 User’s manual

21

IDE Configuration
Serial-ATA (SATA)
Enable / disable Serial ATA.
SATA Test Mode
Test Mode enable / disable.
SATA Speed Support
SATA Speed Support Gen1 or Gen2.
SATA ODD PortSATA ODD is Port0 or Port1.
SATA Mode
Select IDE / AHCI
Serial-ATA Port 0
Enable / Disable Serial ATA Port 0.
SATA Port0 HotPlug
Enable / Disable SATA Port0 HotPlug.
Serial-ATA Port 1
Enable / Disable Serial ATA Port 1.
SATA Port0 HotPlug
Enable / Disable SATA Port1 HotPlug
Embedded Controller Configuration
Lock key click time
Number of seconds to press lock key to (un)active.
1~10 seconds mean press hold time.
Locked Status Hold Time
Number of minutes to keep lock status. 0 means lock always.
1~100 minutes mean lock hold time.
Power key
Enable / Disable power key.
Mute key
Enable / Disable Mute key.
22

WMP-153 User’s manualVolume Up/Dn keys
Enable / Disable Vol+/Vol- key.
LCD Brightness Up/Dn keys
Enable / Disable LCD Brightness up/down key.
LCD Backlight On / Off key
Enable / Disable LCD Backlight key.
Lock key
Enable / Disable Lock key.
Webcam key
Enable / Disable webcam key.
Fn keyEnable/Disable Function key
Miscellaneous Configuration
PCI Express Dynamic Clock Gating
Enable / Disable PCIE Dynamic Clock Gating.
Show/hide hidden items
For debug only.
Show / hide hidden items.
Info Report Configuration
Post Report
Post Repost Support Enabled / Disabled.
Info Error Message
Info Error Message Support Enabled / Disabled.
Summary Screen
Summary Screen Support Enabled / Disabled.
USB Configuration
USB Support
USB Support Parameters.

WMP-153 User’s manual23Legacy USB Support
Enables Legacy USB support. AUTO option disables legacy
support if no USB devices are connected. DISABLE option willkeep USB devices available only for EFI applications
XHCI Hand-off
Enable / Disable XHCI Controller Legacy support.
USB Mass Storage Driver Support
Enable/Disable USB Mass Storage Driver Support.

24

WMP-153 User’s manualChipset

North Bridge
Config Intel IGD Settings.
South Bridge
High Precision Timer
Enable or Disable the High Precision Event Timer.
Audio Controller
Control Detection of the Azalia device.
Disabled = Azalia will be unconditionally disabled.
Enable = Azalia will be unconditionally Enabled.
Auto = Azalia will be enabled if present disabled otherwise.
Restore AC Power Loss
Select AC power state when power is re-applied after a power
failure.

WMP-153 User’s manual25Security

Administrator Password
Set Administrator Password.
User Password
Set user Password.
P0：TOSHIBA MQ01
HDD Security Configuration for selected drive.

26

WMP-153 User’s manualBoot

Setup Prompt Timeout
Number of seconds to wait for setup activation key.
65535(0xFFFF) means indefinite waiting.
Bootup Numlock State
Selects the keyboard NumLock state.
PXE Boot
PXE Network Boot Enable / Disable.
OS Selection
OS Selection.
Quiet Boot
Enable or disables Quiet Boot option.
Boot Option #1
Sets the system boot order.
Boot Option #2
Sets the system boot order.

WMP-153 User’s manual27

Save & Exit

Save Changes and Exit
Exit system setup after saving the changes.
Discard Changes and Exit
Exit system setup without saving any changes.
Save Changes and Reset
Reset the system after saving the changes.
Discard Changes and Reset
Reset system setup without saving the changes.
Save Changes
Save the changes done so far to any of setup options.
Discard Changes
Discard the changes done so far to any of setup options.
Restore Defaults
Restore/load default values for all the setup options.
28

WMP-153 User’s manualSave as User Defaults
Save the changes done so far as User Defaults.
Restore User Defaults
Restore the User Defaults to all the setup options.
Boot Override
To override the boot device.
Launch EFI Shell From filesystem device
Attempts to Launch EFI Shell application (Shell.efi) from one of
the available filesystem devices.
Reset System with ME disable ModeMEUD000
ME will runs into the temporary disable mode, Ignore if ME
Ignition FWMEUD001.

WMP-153 User’s manual29

APPENDIX

Appendix
A. Jumper settings and Connectors

This appendix gives the definitions and shows the positions ofjumpers, headers and connectors. All of the configuration
jumpers on WMP-153 are in the proper position.
Note: Some of jumpers or connectors will be removed
base on system configuration.

Jumper and Connector Definition Block30

WMP-153 User’s manual

APPENDIX

JP1 – Backlight Adjust
Description
analog Inverter
PWM Inverter

Jumper Setting
1-2 (default)
2-3

JP2 – Backlight Voltage Selection
Description
4 wire5 wire8 wireJumper Setting
1-2, 3-4, 5-6, 7-8, 9-10
3-4, 5-6, 7-8, 9-10 (default)
1-2

JP3 – Touch Panel Type Selection
Description
3M type
ELO type

Jumper Setting
1-2, 3-4 (default)
5-6,7-8

JP4 – LVDS Power Selection
Description+5VS (for 17"/19"/21.5”)+3.3VS (for 10"/12"/15")

Jumper Setting
1-2,3-4 (default)
5-6,7-8

JP5 – Panel Type Selection
GPIO3
GPIO2
GPIO1
GPIO0

0
JP5(1-2) short
JP5(3-4) short
JP5(5-6) short
JP5(7-8) short1
JP5(1-2) open
JP5(3-4) open
JP5(5-6) open
JP5(7-8) open

WMP-153 User’s manual31

APPENDIX

JP6 – CMOS Clear
Description
Normal OpenCMOS Clear

Jumper Setting
1-2 (default)
2-3

JP8,JP9 – Support Y cable with DVI+VGA
Description
DVI
DVI+VGA

Jumper Setting
JP8
Open
Short

Description
DVI
DVI+VGA

JP10 – COM4 Power Selection
Description
+5VS
+12VS

32

Jumper Setting
2-3(default)
1-2

WMP-153 User’s manual

APPENDIX

Connector Definition
PJ1 /PJ2 – HDD Power Connector

Pin #
1
2
3
4

Signal Description+12VS
GroundGround+5VS

PJ4 – Power Jack Connector

Pin #
1
2
3
4
5

Signal Description
DC In
DC In
GroundGroundGround

J1 – LCD Inverter Interface
Pin1

WMP-153 User’s manual33

APPENDIX

Pin #
1
2
3
4
5
6

Signal Description
+12VS
+12VS
Backlight Adjust
Backlight Enable
GroundGroundJ2 – Internal Webcam USB 2.0 Pin Header
Pin1
Pin #
1
2
3
4
5
6

Signal Description
+5VSB
+5VSB
Data Data +GroundGroundJ3 / J11 – DDR3 SO-DIMM Interface
J3 H9.2 STD
J11 H9.2 REV
Pin
1
3
5
7
9
11
13
15
17
19
21
23
25
27
29

SymbolVREFDQ
VSS
DQ0
DQ1
VSS
DM0
VSS
DQ2
DQ3
VSS
DQ8
DQ9
VSS
DQS1#
DQS1

34

Pin
69
71
73
75
77
79
81
83
85
87
89
91
93
95
97

Symbol
DQ27
VSS
CKE0
VDD
NC
BA2
VDD
A12
A9
VDD
A8
A5
VDD
A3
A1

Pin
137
139
141
143
145
147
149
151
153
155
157
159
161
163
165

Symbol
DQS4
VSS
DQ34
DQ35
VSS
DQ40
DQ41
VSS
DM5
VSS
DQ42
DQ43
VSS
DQ48
DQ49

Pin
2
4
6
8
10
12
14
16
18
20
22
24
26
28
30

Symbol
VSS
DQ4
DQ5
VSS
DQS0#
DQS0
VSS
DQ6
DQ7
VSS
DQ12
DQ13
VSS
DM1
RESET#

WMP-153 User’s manual

Pin
70
72
74
76
78
80
82
84
86
88
90
92
94
96
98

Symbol
DQ31
VSS
NC
VDD
NC
NF/A14
VDD
A11
A7
VDD
A6
A4
VDD
A2
A0

Pin
138
140
142
144
146
148
150
152
154
156
158
160
162
164
166

Symbol
VSS
DQ38
DQ39
VSS
DQ44
DQ45
VSS
DQS5#
DQS5
VSS
DQ46
DQ47
VSS
DQ52
DQ53

APPENDIX
31
33
35
37
39
41
43
45
47
49
51
53
55
57
59
61
63

VSS
DQ10
DQ11
VSS
DQ16
DQ17
VSS
DQS2#
DQS2
VSS
DQ18
DQ19
VSS
DQ24
DQ25
VSS
DM3

99
101
103
105
107
109
111
113
115
117
119
121
123
125
127
129
131

VDD
CK0
CK0#
VDD
A10
BA0
VDD
WE#
CAS#
VDD
A13
NC
VDD
NC
VSS
DQ32
DQ33

167
169
171
173
175
177
179
181
183
185
187
189
191
193
195
197
199

65
67

VSS
DQ26

133
135

VSS
DQS4#

201
203

VSS
DQS6#
DQS6
VSS
DQ50
DQ51
VSS
DQ56
DQ57
VSS
DM7
VSS
DQ58
DQ59
VSS
SA0
VDDSP
D
SA1
VTT

32
34
36
38
40
42
44
46
48
50
52
54
56
58
60
62
64

VSS
DQ14
DQ15
VSS
DQ20
DQ21
VSS
DM2
VSS
DQ22
DQ23
VSS
DQ28
DQ29
VSS
DQ3#
DQ3

100
102
104
106
108
110
112
114
116
118
120
122
124
126
128
130
132

VDD
CK1
CK1#
VDD
BA1
RAS#
VDD
S0#
ODT0
VDD
NC
NC
VDD
VREFCA
VSS
DQ36
DQ37

168
170
172
174
176
178
180
182
184
186
188
190
192
194
196
198
200

VSS
DM6
VSS
DQ54
DQ55
VSS
DQ60
DQ61
VSS
DQS7#
DQS7
VSS
DQ62
DQ63
VSS
EVENT#
SDA

66
68

VSS
DQ30

134
136

VSS
DM4

202
204

SCL
VTT

J4 – LVDS Interface

Pin #
1
3
5
7
9
11
13
15
17
19
21
23
25
27
29
31

Signal Description+LCD (+5V/ +3.3V)+LCD (+5V/ +3.3V)
GroundGroundA_RxIn0A_RxIn0+
GroundA_RxIn1A_RxIn1+
GroundA_RxIn2A_RxIn2+
GroundA_CKINA_CKIN+
Ground

Pin #
2
4
6
8
10
12
14
16
18
20
22
24
26
28
30
32

Signal Description+LCD (+5V/ +3.3V)+LCD (+5V/ +3.3V)
GroundGroundB_RxIn0B_RxIn0+
GroundB_RxIn1B_RxIn1+
GroundB_RxIn2B_RxIn2+
GroundB_CKINB_CKIN+
GroundWMP-153 User’s manual

35

APPENDIX

33
35
37
39

A_RxIn3A_RxIn3+
GroundGround

34
36
38
40

B_RxIn3B_RxIn3+
GroundGround

J5 – Res Touch Panel Interface
9

1

Pin #
1
2
3
4
5
6
7
8
9

8-wireUL(X+)UR(Y+)N/A
LR(X-)
LL(Y-)
X+_DRIVE
Y+_DRIVE
X-_DRIVE
Y-_DRIVE

Signal Description
4-wireUL(X+)UR(Y+)N/A
LR(X-)
LL(Y-)
N/A
N/A
N/A
N/A

J6 / J12 – Mini PCI Express Socket

36

WMP-153 User’s manual

5-wireUL(X+)UR(Y+)PROBE
LR(X-)
LL(Y-)
N/A
N/A
N/A
N/A

APPENDIX

Pin #
1
3
5
7
9
11
13
15
17
19
21
23
25
27
29
31
33
35
37
39
41
43
45
47
49
51

Signal Description
WAKE#
Reserved
Reserved
CLKREQ#GND
REFCLKREFCLK+
GNDReserved
Reserved
GND
PERn0
PERp0
GND
GND
PETn0
PETp0
GND
GND
+3.3VSB+3.3VSB
GND
CL_CLK
CL_ DATA
Controller Link RST#
Reserved

Pin #
2
4
6
8
10
12
14
16
18
20
22
24
26
28
30
32
34
36
38
40
42
44
46
48
50
52

Signal Description+3.3VSB
GND+1.5VS
Reserved
Reserved
Reserved
Reserved
Reserved
GND
Reserved
PERST#+3.3VSB
GND+1.5VS
SMB_CLKSMB_DATAGND
USB_DUSB_D+
GND
Reserved
Reserved
Reserved+1.5VS
GND+3.3VSB

Pin #
5
6
7
8

Signal Description
SI
SCLK
HOLD#
VDD

J7 – BIOS Socket

Pin #
1
2
3
4

Signal Description
CS#
SO
WP#
VSS

WMP-153 User’s manual37

APPENDIX

J8 – PCT Touch

Pin #
1
2
3
4
5
6

Signal Description
+5VSB
+5VSB
Data Data +GroundGround

J9/J14/J15 – Internal USB 2.0 Pin Header (co-layout)
Pin1
Pin #
1
2
3
4
5
6

Signal Description
+5VSB
+5VSB
Data Data +GroundGround

J16– Power / HDD LED

38

WMP-153 User’s manual

APPENDIX

Pin #
1
2
3
4

Signal Description
SATA_LED#+3.3VSB+3.3VSBPWR_LED#J17 – For JTAG

Pin #
1
3
5
7
9

Signal Description
TRST#
TMS
TDI
TCK
TDO

Pin #
2
4
6
8
10

Signal Description+3.3VSB
RDY#
GND
GND
GND

Pin #
2
4
6
8
10
12
14
16

Signal Description
PLT reset#
SERIRQ+3.3VS+5VSB
CLKRUN#SMB CLKSMB DATA+3.3VSB

J18 –TPM / ID-394

Pin #
1
3
5
7
9
11
13
15

Signal Description
LPC AD0
LPC AD1
LPC AD2
LPC AD3
LPC Frame#
Debug CLKGND
SUS_STAT#WMP-153 User’s manual39

APPENDIX

J19 / J21 – Standard SATA Interface

Pin #
1
2
3
4
5
6
7

Signal Description
GroundTx+
TxGround
RxRx+
Ground

J23 – PS2 KB/MS

Pin #
1
2
3
4
5
6

40

Signal Description
KBDATAMSDATA
Ground
+5VSB
KBCLK
MSCLK

WMP-153 User’s manual

APPENDIX

J24 – EC Reset

Pin #
1
2

Signal Description
EC_RESET#GND

J25 – Battery Socket

Pin #
1
2

Signal Description
RTC +3.3V
GND

J27 – GPIO Connect

Pin #
1
3
5
7
9
11

Signal Description
GEN_GPO1
GEN_GPO2
GEN_GPO3
GEN_GPO4
+5V
GND

Pin #
2
4
6
8
10
12

Signal Description
GEN_GPI1
GEN_GPI2
GEN_GPI3
GEN_GPI4+5V
GND

WMP-153 User’s manual41

APPENDIX

J28 –CAP Front Bezel Button Light Sensor Connect

Pin #
1
2
3
4
5
6
7
8
9

Signal Description+5VSB+3.3VSB
KP_SCLKP_SDAPWR_LED#
KP_P_LED
SATA_LED#
GND
GND

J29 –Light Sensor Connect

Pin #
1
2
3
4
5
6
7
8
9

42

Signal Description+5VSB+3.3VSB
KP_SCLKP_SDAPWR_LED#
KP_P_LED
SATA_LED#
GND
GND

WMP-153 User’s manual

APPENDIX

J30 – Internal COM4 Serial Port

Pin #
1
3
5
7
9

Signal Description
232_DCD#
232_SIN
232_SOUT
232_DTR#
GND

Pin #
2
4
6
8
10

Signal Description
232_DSR#
232_RTS#
232_CTS#
232_RI#+5VS / +12VS

Pin #
2
4
6
8
10

Signal Description
DSR#
RTS#
CTS#
RI#
+5VS

J31 – Internal COM4 TTL

Pin #
1
3
5
7
9

Signal Description
DCD#
SIN
SOUT
DTR#
GND

J32 – External USB 3.0 PortWMP-153 User’s manual

43

APPENDIX

Pin #
1
2
3
4
5
6
7
8
9

Signal Description+5V
DataData+
GND
SSRXSSRX+GND
SSTXSSTX+

J33, J42 – RIGHT / LEFT CH for Speaker

Pin #
1
2

Signal Description
J33 (RIGHT CH)
J42 (LEFT CH)
ROUT+
LOUT+
ROUTLOUT-

J34 –External RJ45 Ethernet Port

Pin #
1
2
3
4
5
6
7
8
44

Signal Description
Data0+
Data0Data1+
Data2+
Data2Data1Data3+
Data3WMP-153 User’s manual

APPENDIX

J35 – Power Switch connect

Pin #
1
2

Signal Description
Power ON
GND

J36,J37,J38 – External USB 2.0 Port

Pin #
1
2
3
4

Signal Description+5V
USB_DUSB_D+
GND

J39 – External DVI-I Connector

Pin #
1
3
5

Signal Description
DVI Data2GND
NC

Pin #
2
4
6

Signal Description
DVI Data2+
NC
DVI SCLWMP-153 User’s manual

45

APPENDIX

7
9
11
13
15
17
19
21
23
C1
C3
C5

DVI SDADVI Data1GND
NC
GND
DVI Data0GND
CRT_DATADVI Clock+
CRT_R
CRT_B
GND

8
10
12
14
16
18
20
22
24
C2
C4
C6

VSYNC
DVI Data1+
NC+5V
Hot Plug Detect
DVI Data0+
CRT_CLKGND
DVI ClockCRT_G
HSYNC
GND

J40 / J41 – External Audio Phone Jack

Pin #
J40
J41

Signal Description
Line Out (stereo) Green
Microphone (stereo) Pink

J43 – Internal COM3 Serial Port

Pin #
1
3
5
7
9

46

Signal Description
232_DCD#
232_SIN
232_SOUT
232_DTR#
GND

Pin #
2
4
6
8
10

Signal Description
232_DSR#
232_RTS#
232_CTS#
232_RI#+5VS

WMP-153 User’s manual

APPENDIX

J44 – Reset Button

Pin #
1
2
3
4

Signal Description
SYS_RESET#
GND
GND
GND

J45 – Reset connector

Pin #
1
2

Signal Description
SYS_RESET#GND

J46 – External COM2 Connector

Pin #
1
3
5
7
9

Signal Description
232_DCD
232_SOUT
GND
232_RTS
232_RI

Pin #
2
4
6
8
10

Signal Description
232_SIN
232_DTR
232_DSR
232_CTS
Not Used

WMP-153 User’s manual47

APPENDIX

J47 – External COM1 Connector

Pin #
1
2
3
4
5
6
7
8
9

48

RS-232Carrier Detect
Receive Data
Transmit Data
Data Terminal Ready
GroundData Set Ready
Request to Send
Clear to Send
Ring Indicator

Signal Description
RS-422
Transmit Data Transmit Data +Receive Data +Receive Data NC
NC
NC
NC
NC

WMP-153 User’s manual

RS-485
Transmit Data Transmit Data +
NC
NC
NC
NC
NC
NC
NC

APPENDIX

B. L type Stand (optional kit)
Key Features and BenefitsSimilar color as WMP series
10 degree tilt down and 30 degree tilt up solution
5,000 times hinge life cycle

Specifications:
Weight Capacity:

Max 10kgs

Monitor Mounting Holes

VESA 75*75mm or 100*100mm

Application using

Desktop standWMP-153 User’s manual49

APPENDIX

C. Battery Pack Specifications (optional kit)
Battery Model133

BP-WMP226 22/3900 SA

Battery Type

Li-ion 2S2P

Minimum Capacity

3900 mAh

Nominal Voltage

7.2 V

Max. Charge Voltage

8.4V

Cut Off Voltage

6.0v

Suggested Charge
Current (Max.)
System Continuous
Discharging Current(Max.)The End of Charge
Condition
Discharge Protection

2A

Charge Protection

OVP/OTP

Self-discharge Rate

10uA ~800 uA

Dimensions

133 x 47 x 21mm

Weight240g max.

Ambient Temperature

0°C ~ +40°C

Storage Temperature-20°C ~ +60°C

Energy28.08Wh

Backup53 W/ 40 min

50

16.6 A

150 mA/min
UVP/OCPWMP-153 User’s manual

APPENDIX

D. How to disable battery when system hang up-WMP-153

1. When system hang up ,press power button 10 seconds to turnoff system . If you can’t power on system by power button.
Please follow below step to reset system.
2. Remove AC power cord, then the battery LED will on.
3. Press volume up key 10 seconds, then all LED will turn on andturn off

4. Plug in AC power cord again, press power button then you canpower on system.

WMP-153 User’s manual51

APPENDIX

E. Scrap Computer Recycling
If the computer equipments need the maintenance or are beyond
repair, we strongly recommended that you should inform us assoon as possible for the suitable solution. For the computers thatare no longer useful or work well, please contact with worldwide
distributors for recycling.
The worldwide distributors show on the following website:
http://www.wincomm.com.tw/contact.aspx
Note:
Follow the national requirement to dispose unit

52

WMP-153 User’s manual