ENPRECISE DISPENSING | IKA PETTE fix, vario and multi

2 | PETTE fix, vario and multi

IKA PETTEPETTE fix, vario and multi/// IT FITS SO COMFORTABLY IN YOUR HAND THAT YOU DON'T WANT TO LET GO/// Simple liquid handling for fixed and variable volumes

It's everything you could ever want for precise dispensing combined in one device: With the PETTE fix, vario andmulti pipettes, IKA combines the most popular features of this laboratory classic. With uncompromisingly accurate
performance and the desired reproducibility, the PETTE is also incredibly comfortable in your hand. Each pipettecomes with three different grip sizes. Whether you're left-handed or right-handed, have big hands or small hands the IKA PETTE always adapts perfectly to your own grip. This makes it ideal for a wide range of applications andfrequent laboratory use.

An ergonomic feel combined with uncompromising precise performance:
the IKA single- and multi-channel pipettes have been devised for use inmany applications typical in a modern laboratory. Color coding makes it
simple to find the right pipette. The fact that they are resistant to impact,
UV and chemicals, ensures the pipettes will provide long-term reliable
dispensing. PETTE fix, vario and multi are fully autoclavable, therefore
ensuring easy sterilization. This guarantees fault-free, precise and safe
operation.

Single-channel pipettes
PETTE fix and vario are available in 18 different sizes, from themicropipette with a volume of 0.1 μl, up to a metering volume of
10 ml. Ten models have a fixed volume, whereas eight are variably adjustable.

Multi-channel pipettes
PETTE multi is available with either 8 or 12 channels for accurate
dispensing in microtiter or deep well plates. In total, there are six
different models available with adjustable volumes from 0.5 to200 μl.

4

12/// FEATURES/// COMPLETE LAB SOLUTIONS

Personalized Application Advice

814/// TECHNICAL DATA/// FAQ

11

15/// ACCESSORIES/// SCOPE OF DELIVERY

You can test all the pipettes yourself at the IKA Application
Center. Our experts are available for questions and various testsof all kinds.

Worldwide Service
GERMAN TECHNOLOGYThe IKA PETTE is 100% designed anddeveloped in Germany.

2 years
warranty**1+1 years after registering at www.ika.com/register,
wearing parts excluded

Choosing an IKA PETTE also means choosing the excellent technical service offered by IKA. Our team is available worldwide foryour service and application needs. Availability of spare parts foryour pipette is guaranteed for 10 years. In addition, we calibrate
your pipette on request as often as you need according to standard (ISO 8655).
Any questions? Our service team is available for your specificrequirements:
00 8000 52457 (00 8000 IKAHELP)
service@ika.de4 | Features

Features | 5

A dose of everything you need
Ergonomic design
All IKA pipettes include three exchangeable grips in different
shapes and materials. Designed for both right-handers andleft-handers, as well as for people with small or large hands,
they are guaranteed to fit comfortably in your hand. The minimal
frictional and spring forces also ensure ergonomically-friendly
dispensing; without any effort. One-handed adjustment of thevolume is possible using the multi-functional knob. Moreover, thetip of the pipette can be replaced regardless of your hand position, while the volume adjustment is securely locked. Together
with its pleasantly smooth surface and low weight, the ergonomic design makes for a particularly convenient work experience.

Award-winning IKA PETTE
The IKA PETTE fix and vario pipettes have received two prizes
already. The Council for Design honored the laboratory classic
with the German Design Award 2020 in the category “Excellent
Product Design - Industry”. In addition, the 78-member panelof experts from International Forum Design GmbH in Hanoverpraised the IKA PETTE’s innovative shape and awarded theworld-renowned iF Design Award 2020.

Always readable
The mechanical volume display shows the set volume in large
numbers and is easy to read no matter how you hold it - even while
dispensing.

Thermally insulated
Designed to last
The tip cone and piston are coated with a high-quality coating*(DLC, diamond like carbon). The IKA PETTE is constructed in afunctional and minimalistic way and consist of only a few, clearlyarranged individual parts. This combination results in extra sturdy
pipettes that deliver reliable results day after day and are very
easy to clean.

* Pipettes with volume up to 200 μl

The housing and grip are thermally separated by an air layer. This
layer of air insulates the pipette so that it does not heat up in yourhand. Thanks to this design, typical volume errors for pipettes are
reduced.

Simple maintenance and adjustment
As many parts as necessary and as few as possible: Thanks to its
sophisticated design, the volume unit with piston and seal can bedisassembled and cleaned in just a few simple steps. If adjustment isnecessary, it can be easily undertaken without the need for tools.
For PETTE multi, each tip cone can even be replaced individually. Thisextends the lifetime of the pipettes.

6 | Features

Features | 7

Convenient color coding

High-tech – only the best for your lab

The different colors of the dispensing heads make it easier to choose theright pipette tip.

PETTE FIXIDENT-NO.

MAX. VOLUME TIP

5 μl

0020011219

10 μl

10 μl

0020011220

10 μl

20 μl

0020011221

200 μl25 μl0020011222

200 μl50 μl0020011223

200 μl100 μl0020011224

200 μl200 μl0020011225

200 μl250 μl0020011226

1000 μl

500 μl0020011227

1000 μl1000 μl0020011228

1000 μl

PETTE VARIOIDENT-NO.

MAX. VOLUME TIP

0.1 – 2 μl

0020011210

10 μl

0.5 – 10 μl

0020011211

10 μl

2 – 20 μl

0020011213

200 μl10 – 100 μl

0020011214

200 μl20 – 200 μl

0020011215

200 μl100 – 1000 μl

0020011216

1000 μl0.5 – 5 ml

0020011217

5 ml

1 – 10 ml

0020011218

10 ml

PETTE MULTIIDENT-NO.

MAX. VOLUME TIP

8 x 0.5 – 10 μl

0020017943

10 μl

8 x 10 – 100 μl

0020017944

200 μl8 x 20 – 200 μl

0020017945

200 μl12 x 0.5 – 10 μl

0020017946

10 μl

12 x 10 – 100 μl

0020017947

200 μl12 x 20 – 200 μl

0020017948

200 μlAt IKA, we believe a common piece of everyday lab equipment,
like pipettes, deserves the attention of our best engineers. Afterall, it's about making work easier for users, covering a wide
range of applications and guaranteeing work that conforms tostandards. One look at the state-of-the-art production of the
PETTE reveals a lot about what awaits you with your new pipette.

Pick-by-light assembly
The IKA pipettes are completely assembled according to the pickby-light procedure. Our order pickers are able to assemble thepipettes quickly and without errors using computer-controlled
eye-catching lights and acknowledgment knobs. The pleasing
result for our customers are hardly any return rates and shortdelivery times.

Calibration by robots"Pit" and "Patty" are the names which our developers havelovingly given to the specially developed robots which calibrate
all assembled pipettes fully automatically and in accordance withstandards. Pit and Patty can independently identify inaccuracies
and analyze faults.
Calibration certificate and quality certificate
You receive your PETTE fix, vario and multi with an individual
quality certificate (QC) and calibration certificate according to ISO
8655.
IKA calibration software
The IKA software for independent planning, execution and documentation of calibration intervals is extremely user-friendly.
IKA calibration service
If you would rather have your pipettes calibrated externally, youare also welcome to send them in to our calibration service.

8 | Technical data

Technical data | 9Technical data
TECHNICAL DATAIKA PETTE fix
5 µl | 10 µl | 20 µl | 25 µl | 50 µl | 100 µl | 200 µl

IKA PETTE fix
250 µl | 500 µl | 1000 µl

TECHNICAL DATAIKA PETTE vario
0.1-2 µl | 0.5-10 | 2-20 µl | 10-100 µl | 20-200 µl

IKA PETTE vario
100-1000 µl | 0.5-5 ml | 1-10 ml

DesignAir displacement piston-operated pipette

Air displacement piston-operated pipette

DesignAir displacement piston-operated pipette

Air displacement piston-operated pipette

Handling

Mechanical

MechanicalHandling

Mechanical

MechanicalVolumeFix

Fix

VolumeVariable

Variable

Number of channels

11

Number of channels

11Colour CodeGray | Gray | Yellow | Yellow | Yellow | Yellow | Yellow

Blue

Colour CodeGray | Gray | Yellow | Yellow | Yellow

Blue | Green | Red

Nominal volume

5 µl | 10 µl | 20 µl | 25 µl | 50 µl | 100 µl | 200 µl

250 µl | 500 µl | 1000 µl

Nominal volume

2 µl | 10 µl | 20 µl | 100 µl | 200 µl

1000 µl | 5000 µl | 10 000 µl

Accuracy with nominal volume±0.06 µl | ±0.12 µl | ±0.2 µl | ±0.25 µl | ±0.35 µl | ±0.6 µl | ±1.2 µl±1.5 µl | ±3 µl | ±6 µl

Volume min.

0.1 µl | 0.5 µl | 2 µl | 10 µl | 20 µl

100 µl | 500 µl | 1000 µl

Accuracy with nominal volume±1.2 % | ±1.2 % | ±1 % | ±1 % | ±0.7 % | ±0.6 % | ±0.6 %±0.6 %

Volume max.

2 µl | 10 µl | 20 µl | 100 µl | 200 µl

1000 µl | 5000 µl | 10 000 µl

Precision with nominal volume±0.03 µl | ±0.06 µl | ±0.06 µl | ±0.075 µl | ±0.15 µl | ±0.2 µl | ±0.4 µl±0.5 µl | ±1 µl | ±2 µl

Increments

0.002 µl | 0.01 µl | 0.02 µl | 0.1 µl | 0.2 µl

1 µl | 5 µl | 10 µl

Precision with nominal volume±0.6 % | ±0.6 % | ±0.3 % | ±0.3 % | ±0.3 % | ±0.2 % | ±0.2 %±0.2 %

Accuracy with nominal volume±0.028 µl | ±0.1 µl | ±0.2 µl | ±0.8 µl | ±1.2 µl±6 µl | ±30 µl | ±60 µl

Certificate of confirmity
according to ISO 8655

Yes

Yes

Accuracy with nominal volume±1.4 % | ±1 % | ±1 % | ±0.8 % | ±0.6 %±0.6 % | ±0.6 % | ±0.6 %Precision with nominal volume±0.014 µl | ±0.04 µl | ±0.06 µl | ±0.2 µl | ±0.4 µl±2 µl | ±7.5 µl | ±15 µl

Autoclavable

Yes

Yes

Precision with nominal volume±0.7 % | ±0.4 % | ±0.3 % | ±0.2 % | ±0.2 %±0.2 % | ±0.15 % | ±0.15 %Dimensions (W x H x D)

30 x 248 x 63 mm

30 x 247 x 63 mm

Accuracy with 50% of nominal volume±0.025 µl | ±0.075 µl | ±0.12 µl | ±0.5 µl | ±1 µl±5 µl | ±30 µl | ±40 µl

Weight0.084 kg | 0.084 kg | 0.085 kg | 0.085 kg | 0.085 kg | 0.085 kg | 0.086 kg

0.083 kgAccuracy with 50% of nominal volume±2.5 % | ±1.5 % | ±1.2 % | ±1 % | ±1 %±1 % | ±1.2 % | ±0.8 %Permissible ambient temperature5 – 40 °C

5 – 40 °C

Precision with 50% of nominal volume±0.015 µl | ±0.04 µl | ±0.06 µl | ±0.15 µl | ±0.3 µl±1 µl | ±6.25 µl | ±10 µl

Permissible relative humidity

90 %

90 %

Precision with 50% of nominal volume±1.5 % | ±0.8 % | ±0.6 % | ±0.3 % | ±0.3 %±0.2 % | ±0.25 % | ±0.2 %Accuracy with 10% of nominal volume±0.024 µl | ±0.025 µl | ±0.1 µl | ±0.3 µl | ±0.5 µl±3 µl | ±12 µl | ±30 µl

Accuracy with 10% of nominal volume±12 % | ±2.5 % | ±5 % | ±3 % | ±2.5 %±3 % | ±2.4 % | ±3 %Precision with 10% of nominal volume±0.012 µl | ±0.018 µl | ±0.03 µl | ±0.1 µl | ±0.14 µl±0.6 µl | ±3 µl | ±6 µl

Precision with 10% of nominal volume±6 % | ±1.8 % | ±1.5 % | ±1 % | ±0.7 %±0.6 % | ±0.6 % | ±0.6 %Certificate of confirmity according to ISO 8655

Yes

Yes

Autoclavable

Yes

Yes

Dimensions (W x H x D)

30 x 248 x 63 mm

30 x 247 x 63 mm | 30 x 219 x 63 mm |30 x 222 x 63 mm

Weight0.089 kg | 0.089 kg | 0.09 kg | 0.09 kg | 0.091 kg

0.088 kg | 0.099 kg | 0.108 kg

The accuracy and precision data listed in the table (acc. to DIN EN ISO 8655) only apply if IKA pipette tips are used.
Technical specifications are subject to change.

10 | Technical data

Accessories | 11

Accessories
To complement the work with PETTE fix, vario and multi pipettes, the following accessories from
IKA are available for your laboratory:

LINEA Pipette rack
Secure and easy accessible mount for frequently used
pipettes.
Ident-No.: 0020018987

RONDA Pipette carousel standHolds up to 6 single- or multi-channel pipettes.
Ident-No.: 0020107134

Pipette tips
The IKA range of pipette tips includes tips with and without
filters. They provide a more accurate sample delivery andincreased reproducibility. Available in sizes 0.1 μl – 10 ml.

Pipette calibration software
IKA provides user-friendly pipette calibration software forplanning, performing and documenting calibration tests.
Ident-No.: 20022141

TECHNICAL DATAIKA PETTE multi
8 x 0.5-10 µl | 10-100 µl | 20-200 µl

IKA PETTE multi
12 x 0.5-10 µl |10-100 µl | 20-200 µl

DesignAir displacement piston-operated pipette

Air displacement piston-operated pipette

Handling

Mechanical

MechanicalVolumeVariable

Variable

Number of channels

812Colour CodeGray | Yellow | Yellow

Gray | Yellow | Yellow

Nominal volume

10 µl | 100 µl | 200 µl

10 µl | 100 µl | 200 µl

Volume min.

0.5 µl | 10 µl | 20 µl

0.5 µl | 10 µl | 20 µl

Volume max.

10 µl | 100 µl | 200 µl

10 µl | 100 µl | 200 µl

Increments

0.01 µl | 0.1 µl | 0.2 µl

0.01 µl | 0.1 µl | 0.2 µl

Accuracy with nominal volume±0.2 µl | ±0.8 µl | ±1.6 µl±0.2 µl | ±0.8 µl | ±1.6 µl

ACCESSORIESDESCRIPTIONIDENT-NO.

LINEA

Pipette rack0020018987

RONDA

Pipette carousel stand0020107134

Tip xs bag

Pipette tip, 10 µl, transparent

0020017816

Accuracy with nominal volume±2 % | ±0.8 % | ±0.8 %±2 % | ±0.8 % | ±0.8 %Precision with nominal volume±0.1 µl | ±0.3 µl | ±0.6 µl±0.1 µl | ±0.3 µl | ±0.6 µl

Precision with nominal volume±1 % | ±0.3 % | ±0.3 %±1 % | ±0.3 % | ±0.3 %Tip xs tray

Pipette tip, 10 µl, transparent

0020017818

Accuracy with 50% of nominal volume±0.2 µl | ±0.5 µl | ±1 µl±0.2 µl | ±0.5 µl | ±1 µl

Tip xs box +Filter tip, 10 µl, sterile, transparent

0020017820

Accuracy with 50% of nominal volume±4 % | ±1 % | ±1 %±4 % | ±1 % | ±1 %Tip s bag

Pipette tip, 200 µl, transparent

0020017822

Tip s tray

Pipette tip, 200 µl, transparent

0020017823

Tip s box +Filter tip, 200 µl, sterile, transparent

0020017824

Tip m bag

Pipette tip, 1000 µl, transparent

0020017826

Precision with 50% of nominal volume±0.1 µl | ±0.4 µl | ±0.65 µl±0.1 µl | ±0.4 µl | ±0.65 µl

Precision with 50% of nominal volume±2 % | ±0.8 % | ±0.65 %±2 % | ±0.8 % | ±0.65 %Accuracy with 10% of nominal volume±0.08 µl | ±0.3 µl | ±0.6 µl±0.08 µl | ±0.3 µl | ±0.6 µl

Tip m box

Pipette tip, 1000 µl, transparent

0020017827

Accuracy with 10% of nominal volume±8 % | ±3 % | ±3 %±8 % | ±3 % | ±3 %Tip m box +Filter tip, 1000 µl, sterile, transparent

0020017828

Precision with 10% of nominal volume±0.05 µl | ±0.2 µl | ±0.25 µl±0.05 µl | ±0.2 µl | ±0.25 µl

Tip l bag

Pipette tip, 5 ml, transparent

0020017829

Precision with 10% of nominal volume±5 % | ±2 % | ±1.25 %±5 % | ±2 % | ±1.25 %Tip l box

Pipette tip, 5 ml, transparent

0020017830

Tip xl bag

Pipette tip, 10 ml, transparent

0020017831

Tip xl box

Pipette tip, 10 ml, transparent

0020017832

IKA Tip 10 µl bag

Pipette tip, 10 µl, transparent

0020114634

Certificate of confirmity according to ISO 8655

Yes

Yes

Autoclavable

Yes

Yes

Dimensions (W x H x D)

83 x 235 x 63 mm

119 x 235 x 63 mm

IKA Tip 10 µl box

Pipette tip, 10 µl, transparent

0020116857

Weight0.135 kg | 0.147 kg | 0.160 kg

0.160 kg | 0.182 kg | 0.195 kg

IKA Tip 200 µl bag

Pipette tip, 200 µl, transparent

0020114635

IKA Tip 200 µl box

Pipette tip, 200 µl, transparent

0020116858

IKA Tip 1000 µl bag

Pipette tip, 1000 µl, transparent

0020114636

IKA Tip 1000 µl box

Pipette tip, 1000 µl, transparent

0020116859

The accuracy and precision data listed in the table (acc. to DIN EN ISO 8655) only apply if IKA pipette tips are used.
Technical specifications are subject to change.

12 | Complete lab solutions

Complete lab solutions | 13

Complete lab solutions

Shakers
IKA offers the perfect laboratory shaker for all mixing and shaking tasks. In addition to the tried and tested orbital and horizontal shakers, this also includes the
incubation shaker.

Our goal is to always offer you not only the highest quality devices, but also perfect laboratory solutions. And whoever uses pipettes for dispensing also needs ...

Highlights:> Keep samples in motion> Reach up to 3000 rpm in continuous operation> Offer digital display, timer function and control via the laboratory software
labworldsoft®
> Have an extensive range of attachments> Are durable and safe

IKA PCR 96
Insert

IKA 0.5 ml
InsertIKA 1.5 / 2.0 ml
Insert

MATRIX Orbital
Ident-No.: 0030000627

Thermoshakers
MATRIX represents the high-performance product family of IKA thermal shakers.
Even the smallest volumes mix reliably and completely in all laboratory applications(i.e. blood samples, pharmaceutical agents, DNA/RNA samples or ELISA assays).
There is no cross-contamination; instead, what you get is optimum mixing. This iswhat the IKA MATRIX thermoshakers are all about:> You can mix, heat, cool or even all at the same time
> They are impressive, with a speed of up to 3000 rpm> Aluminum die cast housing ensures secure positioning and excellent stability> Variable inserts broaden the scope of application

Dispersers

Loopster digital
Ident-No.: 0004016000

Trayster basic
Ident-No.: 0004005000

ROCKER 3D digital
Ident-No.: 0004001000

VORTEX 3
Ident-No.: 0003340000

KS 3000 ic control
Ident-No.: 0003940100

The IKA UTTD power control disperser assures the best stirring, dispersing, grinding and homogenization results and has been devised in particular for the safe
processing of infectious, toxic or odor-intensive samples. A particularly powerful
motor gives it its name.
Its strengths at a glance:

ULTRA-TURRAX®
Tube Drive Power control
Ident-No.: 0025005981> Disposable sample tubes systematically prevent cross-contamination> T ime can be preprogrammed and frequently used parameters can be stored in
the library> The direction of rotation can be reversed> T he turbo button brings even more intensity when mixing, dispersing or grinding14 | FAQ

Scope of delivery | 15

FAQ
1. Is every pipette individually calibrated upon delivery?
Yes, every pipette is reliably calibrated at IKA using state-of-theart robotics. The calibration complies with EN ISO 8655.

Scope of delivery
6. Which liquids are ideally used for reverse pipetting?
Reverse pipetting is most suitable for liquids with high vapor
pressure, high viscosity and a foaming tendency. It is also ideal
for very small volumes. On the other hand, forward pipetting
is used in aqueous solutions such as buffer solutions or dilute
solutions.

2. Can I also calibrate the pipette later?
Yes, calibration tests for PETTE fix, vario and multi are extremelyuser-friendly with the IKA IPCS Pipette calibration software.

3. Can I also have my pipette calibrated by IKA?
We are happy to take care of the calibration of your pipettes
according to standards. Our service team is available to help youwith any questions you may have.
By e-mail: service@ika.de
By phone: 00 8000 4524357

4. Are tools needed for the readjustment?
No, PETTE fix, vario and multi are designed in a way that you do
not need any tools for readjustment.

5. Do tips from other manufacturers also fit onto IKA pipettes?
All models of the IKA PETTE are compatible with pipette tips frommost manufacturers.

7. How to sterilize IKA PETTE fix, vario or multi?
IKA pipettes are fully autoclavable by heat sterilization under thefollowing conditions: 121 °C, 2 bar for 20 minutes. Thanks to its
simple design, the pipette can be cleaned and autoclaved evenwhen disassembled. Alternatively, methods using UV radiation
can be used.

Pipette3 grips

Inert grease

Everything for your Lab

BIOREACTORSMIXINGHEATING / COOLING /TEMPERING

LIQUID HANDLINGMEASURING VISCOSITYVACUUM TECHNOLOGY

CRUSHING

LABORATORY SOFTWARESEPARATIONELECTROCHEMISTRY
FLOW CHEMISTRY

REACTOR SYSTEMSCALORIMETRY

IKA-Werke GmbH & Co. KG
Janke & Kunkel-Straße 10,79219 Staufen, Deutschland+49 7633 831-0 / sales@ika.de

IKAworldwide // #lookattheblue

Stay up to date:
www.ika.com/
newsletter

Technical specifications are subject tochange without notice.
Information on delivery not binding

202409_Pette_Brochure_IWS_EN_94000660

IKA is your reliable partner in laboratory, analysis and process technology. When it comes to applications in the areas of stirring, m­ ixing,
tempering, distilling or grinding, market leaders rely on our proven products and technology. We are continually working to further develop our portfolio according to the needs of our customers. Building on this, we network our products and services to create h­ olistic,
application-oriented solutions and enable our customers to achieve the best possible results and step into the digital age.