K-ASSAY 

KAMIYA BIOMEDICAL COMPANY

KAMIYA BIOMEDICAL COMPANY

Pig Fibrinogen ELISA
For the quantitative determination of Fibrinogenin pig biological samples

Cat. No. KT-484

For Research Use Only.

1Rev. 9925484

K-ASSAY 

KAMIYA BIOMEDICAL COMPANYPRODUCT INFORMATION

Pig Fibrinogen ELISA
Cat. No. KT-484
INTENDED USE
The Pig Fibrinogen ELISA is a highly sensitive two-site enzyme-linked immunoassay (ELISA) for the quantitative
determination of Fibrinogen in pig biological samples. For research use only.

INTRODUCTION
Soluble Fibrinogen (FIB) circulates in the blood and provides the material from which the insoluble fibrin clot is formed
during blood coagulation. Fibrinogen is an acute phase reactant that may be a useful marker for infection andinflammation. This ELISA kit can be used to measure Fibrinogen in biological samples.

PRINCIPLE
The principle of the double antibody sandwich ELISA is represented in Figure 1. In this assay the Fibrinogen present in
the sample reacts with the anti-Fibrinogen antibodies which have been adsorbed to the surface of polystyrene microtiterwells. After the removal of unbound sample proteins by washing, anti-Fibrinogen antibodies conjugated with horseradish
peroxidase (HRP), are added. These enzyme-labeled antibodies form complexes with the previously bound Fibrinogen.
Following another washing step, the enzyme bound to the immunosorbent is assayed by the addition of a chromogenic
substrate, 3,3’,5,5’-tetramethylbenzidine (TMB). The quantity of bound enzyme varies directly with the concentration ofFibrinogen in the sample tested; thus, the absorbance, at 450 nm, is a measure of the concentration of Fibrinogen in thetest sample. The quantity of Fibrinogen in the test sample can be interpolated from the calibration curve constructed fromthe calibrators, and corrected for sample dilution.

Figure 1.
Anti-Fibrinogen Antibodies Bound To Solid Phase│
Calibrators and Samples Added│
Fibrinogen * Anti-Fibrinogen Complexes Formed│
Unbound Sample Proteins Removed│
Anti-Fibrinogen-HRP Conjugate Added│
Anti-Fibrinogen-HRP * Fibrinogen * Anti-Fibrinogen
Complexes Formed│
Unbound Anti-Fibrinogen-HRP Removed│
TMB Substrate Added│
Determine Bound Enzyme Activity

COMPONENTS
1. Diluent Concentrate
One bottle containing 50 mL of a 5X concentrated diluent running buffer.
2

Rev. 9925484

KAMIYA BIOMEDICAL COMPANY

2. Wash Solution Concentrate
One bottle containing 50 mL of a 20X concentrated wash solution.
3. Enzyme-Antibody Conjugate Concentrate
One vial containing 150 µL of a 100X concentrated affinity-purified anti-pig Fibrinogen antibody conjugated with HRP ina stabilizing buffer.
4. TMB Substrate Solution
One vial containing 12 mL of TMB and hydrogen peroxide in citric acid buffer at pH 3.3.
5. Stop Solution
One vial containing 12 mL of 0.3 M sulfuric acid. WARNING: Avoid contact with skin.
6. Microtiter Plate
Twelve removable eight-well strips in well holder frame. Wells are coated with affinity-purified anti-pig Fibrinogen.
7. Pig Fibrinogen Calibrator
One vial containing a Pig Fibrinogen Calibrator.

MATERIALS REQUIRED BUT NOT PROVIDED
••
•
•
•
•
•

Test tubes
Precision pipette (2 µL to 200 µL) for making and dispensing dilutions.
Microplate washer/aspirator
Distilled or de-ionized H2O
Microplate reader
Assorted glassware for the preparation of reagents and buffer solutions
Timer

PRECAUTIONS
1. Read the instructions carefully before beginning the assay.
2. This kit is for research use only.
3. Great care has been taken to ensure the quality and reliability of this product. However, it is possible that in certain
cases, unusual results may be obtained due to high levels of interfering factors.
4. No additives or preservatives are necessary to maintain the integrity of the specimen. Avoid azide contamination.
5. Azide and thimerosal at concentrations higher than 0.1% inhibit the enzyme reaction.
6. Other precautions: Do not interchange kit components from different lots.
 Do not use kit components beyond the expiration date.
 Protect reagents from direct sunlight.
 Do not pipette by mouth.
 Do not eat, drink, smoke or apply cosmetics where reagents are used.
 Avoid all contact with the reagents by using gloves.
 Stop solution contains diluted sulfuric acid. Irritation to eyes and skin is possible. Flush with water aftercontact.

REAGENT PREPARATION
1. Diluent Concentrate
The Diluent solution supplied is a 5X concentrate and must be diluted 1:5 with distilled or de-ionized water.
2. Wash Solution Concentrate
The Wash Solution supplied is a 20X concentrate and must be diluted 1:20 with distilled or de-ionized water. Crystal
formation in the concentrate is not uncommon when storage temperatures are low. Warming of the concentrate to30-35°C before dilution can dissolve crystals.
3. Enzyme-Antibody Conjugate Concentrate
Calculate the required amount of working conjugate solution for each microtiter plate test strip by adding 10 µL
Enzyme-Antibody Conjugate to 990 µL of 1X Diluent for each test strip to be used for testing. Mix uniformly, but gently.
Avoid foaming.
4. TMB Substrate Solution
Ready to use as supplied.
3

Rev. 9925484

KAMIYA BIOMEDICAL COMPANY

5. Stop Solution
Ready to use as supplied.
6. Microtiter Plate
Ready to use as supplied. Unseal microtiter pouch and remove plate from pouch. Remove all strips and wells that will
not be used in the assay and place back in pouch and re-seal along with desiccant.
7. Pig Fibrinogen Calibrator
The Pig Fibrinogen Calibrator should be aliquoted out and stored frozen. It is at a concentration of 18.42 µg/mL andneeds to be diluted in 1X diluent according to the chart below for each run. Mix well between each step. Avoid foaming.
Calibrator

Concentration(ng/mL)

6400543210200100502512.50Calibrator Volume added
to 1X Diluent
20 µL Pig Fibrinogen
Calibrator
250 µL Calibrator 6250 µL Calibrator 5250 µL Calibrator 4250 µL Calibrator 3250 µL Calibrator 2

Volume of 1X Diluent
901 µL250 µL250 µL250 µL250 µL250 µL500 µLSTORAGE AND STABILITY
1. Complete Kit
The expiration date for the kit is stated on the outer label. The recommended storage temperature is 4°C. Note: Seelong-term storage recommendations below for the Pig Fibrinogen Calibrator.
2. Diluent
The 5X Diluent Concentrate is stable until the expiration date. The 1X working solution is stable for at least one weekfrom the date of preparation. Both solutions should be stored at 4°C.
3. Wash Solution
The 20X Wash Solution Concentrate is stable until the expiration date. The 1X working solution is stable for at least
one week from the date of preparation. Both solutions can be stored at room temperature (RT, 16-25°C) or at 4°C.
4. Enzyme-Antibody Conjugate
Undiluted horseradish peroxidase anti-Fibrinogen conjugate should be stored at 4°C and diluted immediately prior touse. The working conjugate solution is stable for up to 1 hour when stored in the dark.
5. TMB Substrate Solution
The TMB Substrate Solution should be stored at 4°C and is stable until the expiration date.
6. Stop Solution
The Stop Solution should be stored at 4°C and is stable until the expiration date.
7. Microtiter Plate
Anti-pig Fibrinogen coated wells are stable until the expiration date and should be stored at 4°C in the sealed foil pouchwith a desiccant pack.
8. Pig Fibrinogen Calibrator
Long Term Storage: Upon receipt, aliquot the calibrator and store them frozen. They will be stable until expiration date.
Short Term Storage: The calibrator is stable for up to 14 days at 4°C. The working calibrator solutions should beprepared immediately prior to use and are stable for up to 8 hours.

INDICATIONS OF INSTABILITY
If the test is performing correctly, the results observed with the calibrator solutions should be within 20% of the expectedvalues.

4Rev. 9925484

KAMIYA BIOMEDICAL COMPANY

SPECIMEN COLLECTION AND HANDLING
Blood should be collected by venipuncture. The serum should be separated from the cells after clot formation bycentrifugation. For plasma samples, blood should be collected into a container with an anticoagulant and then centrifuged.
Care should be taken to minimize hemolysis, excessive hemolysis can impact your results. Assay immediately or aliquot
and store samples at -20°C. Avoid repeated freeze-thaw cycles.
For any sample that might contain pathogens, care must be taken to prevent contact with open wounds. No additives orpreservatives are necessary to maintain the integrity of the specimen. Avoid azide contamination.

ASSAY PROTOCOL
Dilution of Samples
Due to the high-sensitive nature of the assay, each test sample should be diluted before use. A 1:100 dilution isappropriate for most serum samples, and a 1:10,000 dilution is appropriate for most plasma samples. For absolute
quantification of samples that yield results outside the range of the calibration curve, a lesser or greater dilution might berequired. If unsure of sample level, a serial dilution with one or two representative samples before running the entire plateis highly recommended.
1. To prepare a 1:100 dilution of sample, transfer 3 µL of sample to 297 µL of 1X Diluent. This gives you a 1:100dilution. Mix thoroughly.
2. To prepare a 1:10,000 dilution of sample, transfer 5 µL of sample to 495 µL of 1X Diluent. This gives you a 1:100dilution. Next, add 5 µL of your 1:100 diluted sample to 495 µL of 1X Diluent. You now have a 1:10,000 dilution ofyour sample. Mix thoroughly at each stage.

Procedure1. Bring all reagents to RT before use.
2. Pipette 100 µL of
Calibrator 0 (0.0 ng/mL) in duplicate
Calibrator 1 (12.5 ng/mL) in duplicate
Calibrator 2 (25 ng/mL) in duplicate
Calibrator 3 (50 ng/mL) in duplicate
Calibrator 4 (100 ng/mL) in duplicate
Calibrator 5 (200 ng/mL) in duplicate
Calibrator 6 (400 ng/mL) in duplicate
3. Pipette 100 µL of sample (in duplicate) into pre-designated wells.
4. Incubate the Microtiter Plate at 22°C (RT) for thirty (30 ± 2) minutes. Keep plate covered and level during incubation.
5. Following incubation, aspirate the contents of the wells.
6. Completely fill each well with appropriately diluted Wash Solution and aspirate. Repeat three times, for a total of fourwashes. If washing manually: completely fill wells with diluted Wash Solution, invert the plate and pour/shake out thecontents in a waste container. Follow this by sharply striking the wells on absorbent paper to remove residual Wash
Solution. Repeat three times for a total of four washes.
7. Pipette 100 µL of appropriately diluted Enzyme-Antibody Conjugate to each well. Incubate at 22°C (RT) for thirty (30 ±2) minutes. Keep plate covered in the dark and level during incubation.
8. Wash and blot the wells as described in Steps 5 and 6.
9. Pipette 100 µL of TMB Substrate Solution into each well.
10. Incubate in the dark at RT for precisely ten (10) minutes.
11. After ten (10) minutes, add 100 µL of Stop Solution to each well.
12. Determine the absorbance at 450 nm of the contents of each well. Calibrate the plate reader to manufacturer’sspecifications.

5Rev. 9925484

KAMIYA BIOMEDICAL COMPANYThe absorbance of the final reaction mixture can be measured up to two hours after the addition of the Stop Solution.
However, good laboratory practice dictates that the measurement be made as soon as possible.

RESULTS
1. Subtract the average background value from the test values for each sample.
2. Using the results observed for the calibrators construct a calibration curve. The appropriate curve fit is that of a fourparameter logistics curve. A second order polynomial (quadratic) or other curve fits may also be used.
3. Interpolate test sample values from the calibration curve. Correct for sample dilution factor to arrive at Fibrinogenconcentration in original sample.

QUALITY CONTROLIn accord with good laboratory practice, the assays for specific Fibrinogen require meticulous quality control. Eachlaboratory should use routine quality control procedures to establish inter- and intra-assay precision and performance
characteristics.

LIMITATION OF THE PROCEDURE
1. Reliable and reproducible results will be obtained when the assay procedure is carried out with a complete
understanding of the information contained in the package insert instructions and with adherence to good laboratory
practice.
2. Factors that might affect the performance of the assay include proper instrument function, cleanliness of glassware,
quality of distilled or de-ionized water, and accuracy of reagent and sample pipettings, washing technique, incubation
time or temperature.

FOR RESEARCH USE ONLY

KAMIYA BIOMEDICAL COMPANY
12779 Gateway Drive, Seattle, WA 98168
Tel: (206) 575-8068 Fax: (206) 575-8094
Email: LifeScience@k-assay.com
www.k-assay.com

6Rev. 9925484