和文は裏面をご覧下さい。

QUICK REFERENCE [Applicable model: SGB, SGBW, SGBL]

No.SGB12JE1 2018.10
Terminal Arrangement

Model Name

SGB, SGBL

SGB□-□□□□ -□-□

SGBW

Circuit Configuration

SGBL only

SGB, SGBL

RS-485 serial communication
Modular jack pin arrangement

Option0: No option needed, 1: Multi-rotation trimmer
2: Moisture-proof treatment
3: Multi-rotation trimmer+Moisture-proof treatment
Power supply
0: 100 to 240 V AC,
1: 24 V AC/DC*2
Output 2*1
Output*2, Output 1*1
Input
Series name
SGB: 1-output, SGBW: 2-outputs
SGBL: 1-output + Serial communication

No. 1
No. 6

No. 1 COM

No. 1
No. 6

No. 3 YB(+)

No. 2NC

No. 4 YA(-)
No. 5NC

No. 6 COM

*1: For SGBW only
*2: For SGB, SGBL only
(For detailed input/output spec, refer to the full Instruction Manual.)

Indication
Number, °C/°F
Indication
Alphabet
Indication
Alphabet

For SGBLKey Operation Flowchart

Characters-10123456789°C°F

A

B

C

DEF

GH

I

JKL

M

NOPQ

RSTUVW

XYZ

Selection Item List
Input type

Decimal point place
No decimal point
4 to 20 mA (Built-in
50 Ω shunt resistor)
1 digit after point
2 digits after point
4 to 20 mA(250Ω shunt resistor)
3 digits after point
Indication unit
4 to 20 mA(50 Ω shunt resistor)
No unit%0 to 20 mA(250Ω shunt resistor)mA
V
0 to 16 mA(62.5Ω shunt resistor)°C
Savesettings
2 to 10 mA(250Ω shunt resistor)
Save
Not save0 to 10 mA(100Ω shunt resistor) Output 1, Output 2 type
4 to 20 mA
1 to 5 mA(100Ω shunt resistor)
0 to 20 mA
0 to 16 mA
0 to 1 mA(1000Ω shunt resistor)
2 to 10 mA
0 to 10 mA
10 to 50 mA(10 Ω shunt resistor)
0 to 10 mV
0 to 100 mV
0 to 10 mV(Input resi. 1 MΩ)
0 to 1 V
0 to 5 V
0 to 50 mV(Input resi. 1 MΩ)
1 to 5 V
0 to 10 V
0 to 60 mV(Input resi. 1 MΩ)-5 to 5 V※1
Output 1,2 decimal point place
0 to 100 mV(Input resi. 1 MΩ)No decimal point
1 digit after point
0 to 1 V(Input resi. 1 MΩ)
2 digits after point
3 digits after point
0 to 5 V(Input resi. 1 MΩ) Output 1,2 Normal/Reverse
1 to 5 V(Input resi. 1 MΩ)-5 to 5 V
(Input resi. 1 MΩ)
0 to 10 V(Input resi. 1 MΩ)-10 to 10 V
(Input resi. 1 MΩ)NormalReverseAuto/Manual
Auto
ManualRatio setting method
Ratio setting
Output value setting(Abbreviation: Input resi.: Input resistance)※1 Not available for the SGBW.

Communication speed
9600 bps
19200 bps
38400 bps
Data bit/Parity
8 bits/No parity
8 bits/Even
8 bits/Odd
Stop bit
1 bit
2 bits

SGBW

Default Display
RUN display
mode 1MODEkey
for 5 sec

DISPkeyRUN display
mode 2Custom display
mode 2

DISP
key

DISP
key

RUN display
mode 3
DISP
key

Custom display
mode 3DISP
key

RUN display
mode 4

Custom display
mode 4

DISP
key

DISP
key

RUN display
mode 5Unlit
display mode

DISP
key

DISP
key

RUN display
mode 6

All unlit
display mode

DISP
key

DISP
key

Custom display
mode 1
DISP
key

Model display
mode
DISP
key

DOWN
key for
3 sec

MODE key, or after
Manual mode auto return
time has elapsed.

Input

Output 1

Output 2

Instrument

setting modesetting modesetting modesetting mode

DISP
keyfor 3 sec

MODEkey

Input type
DISP
keyDecimal
point place
DISP
key
Output 0%value
DISP
key
Output100% value
DISP
key
Indication
unit
DISP
key
Save
settings
DISP
key

Default Display

Manual mode

Display Mode•
: Available only for the SGBW.
• For the details of the display mode, refer to the full Instruction Manual.

Default Display• If the MODE and DISP keys (in that order) are pressed together forapprox. 3 seconds in any display mode, the display mode will become theDefault Display.
• Once the Default Display is set, the DISP key will be in lock status.
• If the DISP key is pressed for approx. 3 seconds, the key lock statuswill be cancelled.
• If the DISP key is pressed while the DISP key is in lock status,
Multi-Display A will indicate.

Setting Mode
• Output 2 setting mode is available only for the SGBW.
• Communication setting mode is available only for the SGBL.
• If the MODE key is pressed for approx. 5 seconds in each setting mode,
the unit will move to the Default Display.

DISPkeyfor 3 sec

MODEkey

DISPkeyfor 3 sec

Output 1
typeOutput 2
type

DISP key
Output 1
decimalpoint place
DISP key
Indication
value atoutput 0%
DISP key
Indication
value atoutput 100%DISP key

DISP key
Output 2
decimalpoint place
DISP key
Indication
value atoutput 0%
DISP key
Indication
value atoutput 100%DISP key

Output 1
low limit
DISP key

Output 2
low limit
DISP key

Output 1
high limit
DISP key

Output 2high limit
DISP key

Output 1
Normal/Reverse

Output 2
Normal/Reverse

DISP key

DISP key

Output 1
ratio
DISP key

Output 2
ratio
DISP key

Output 1
bias
DISP key

Output 2
bias
DISP key

Input point
1
DISP key
Output 1
output value
1
DISP key

Input point
1
DISP key
Output 2
output value
1
DISP key

Input point
2
DISP key
Output 1
output value
2
DISP key

Input point
2
DISP key
Output 2
outputvalue 2
DISP key

Savesettings
DISP key

Savesettings
DISP key

Default Display

Default Display

MODEkey

DISPkeyfor 3 sec

Communication

MODEkeysetting mode
DISP
key
for 3 sec

MODE
key

Customdisplaysetting mode
DISP
key
for 3 sec

Filter time
constant

Instrument
numberMulti-Display
A

DISP
key

DISP
key

DISP
key

Sensorcorrection

Communication
speed

Multi-Display
B

DISP
key

DISP
key

DISP
key

Indicationtime

Data bit/
ParitySavesettings

DISP
key

DISP
key

DISP
key

Auto/Manual
DISP
key
Ratio setting
methodDISP
key
Manual mode
auto return time

Stop bit
DISP
key
Responsedelay time
DISP
key
Save
settings

DISP
key

DISP
key

Save
settings

Default DisplayDISP
key

Default Display

Default Display

MODEkey