Level 1™ H-1200 Fast Flow Fluid Warmerwith Integrated Air Detector/Clamp
Performance You’ll Warm Up To

Standard of Care for Rapid Infusion
For patients with acute circulatory failure, rapid infusion of crystalloid and/or blood products can be a life-saving
intervention and is first-line treatment for acute resuscitation of patients with shock.1 The Level 1 Fast Flow Fluid
Warmer provides a rapid flow of warmed fluids, such as crystalloid or blood product (including red blood cells)
at a maximum flow rate of up to 1400 ml/min.

Warm Recirculating
Solution

The Level 1 Fast Flow Fluid Warmer delivers a rapid infusion of warm fluids and does so while providing an extra
level of protection with an integrated air detector/clamp. If air is detected in the line, the flow is automatically
stopped; the air is quickly removed, and flow restored, without disconnecting from the patient or requiring theadministration set be changed.

Automated Pressure
Chambers> Mounted pressure chambers allow forfast and easy bag changes> Accommodate standard blood andcrystalloid bags
> On/Off toggle switch automaticallypressurizes chambers to a constantpressure of 300 mm Hg

Built-In Safety Features

Air Detector/Clamp> Air detector/clamp automatically stops
flow if air is present, and removes it
without disconnecting from the patient> Operating Mode LED lights green when
the fluid warmer is on, the disposable
is properly installed, and the air
detector/clamp door is closed> Displays reservoir solution and notifies
user if volume is low or is over theacceptable temperature for safe use> Notifies user if disposable set is notproperly installed

Heat Exchanger

Easy to Use> Counter-current recirculating solution
maintained at 107.6°F/42°C provides
effective and stable heating> Disposables are simple to insert andremove, with numbered blocks on
the warmer> Gas vent/filter assembly can be changed
without replacing the entire disposable> Durable surface makes it easy to clean> Large roller wheels allow for easytransport within facility

1 When Minutes Matter: Rapid Infusion in Emergency Care, Journal Article, 2021, Piehl, Park.

> Latex-free> Sterile fluid path
> Designed for single-patient use

Aluminum,Counter-Current
Heat Exchanger> Check Tubing Mode LED lights yellowand an audible alarm sounds when thepatient line tubing has not been
properly placed in the air detector/
clamp slot and/or when the clamp slot
door is not correctly closed> Air detector/clamp mode LED lights red
and an audible alarm sounds when air isdetected in the gas vent/filter assembly.
The patient line is automaticallyclamped, preventing fluid from going
into the patient; it then unclamps andresumes operation once all air isremoved through the gas vent. The line
is not disconnected from the patientHigh Performance Disposable
Administration Sets

D-100 Disposable Administration Set

Level 1 H-31B
Air Detector/Clamp
Full Range of Disposable Administration Sets
and Accessories

Ordering information
Level 1 H-1200 Fast Flow Fluid Warmer with integrated Air Detector/Clamp
Product CodeDescription

Units/Case

H-1200-EN-115V-US

Level 1 Fast Flow Fluid Warmer with integrated Air Detector/Clamp

1H-31-EN

Level 1 Air Detector/Clamp

1Disposable Normothermic IV Administration Sets
Product CodeDescription

Units/Case

D-60HL

Maximum normothermic flow rate: 530 mL/min

10D-70Maximum normothermic flow rate: 675 mL/min

20D-100

Maximum normothermic flow rate: 950 mL/min

10D-300

Maximum normothermic flow rate: 1,100 mL/min

10Product CodeDescription

Units/Case

F-10

Gas Vent/Filter Assembly (Replacement for D-60HL, D-70, D-100)

20F-30Gas Vent/Filter Assembly (Replacement for D-300)

20

PF-1

Fast Flow Pre-Filter (use one per unit of blood)

60

SC-3

Fast Flow 3-Way Stopcock (3.0 mm/9Fr inner diameter)

40

X-36

Fast Flow Extension Line (91.4 cm/36 in long)40

80-04-003

O-Ring Kit

180-01-040

DSTA-40 FFW Tempcheck

1Y-INJ

Fast Flow Extension Set with Injection Site (20.3 cm/8 in long)

40

Y-30

High-flow Y-Extension Set

40

Accessories© 2022 ICU Medical Inc.

| P22-5783

For safe and proper use, refer to the Instructions for Use.

800.824.7890 | www.icumed.com