EN

OPTIMIZING CHEMICAL PROCESSES | EasySyn Reactor Systems

2 | EasySyn

EasySyn/// The reactor system for chemical synthesis
We are expanding the portfolio of IKA reactor systems: EasySyn is optimized for organic andaqueous syntheses. The controllable and reproducible reaction conditions make EasySyn ideal
for applications in research and development, the chemical industry or materials research.
In the pre-production stage, EasySyn offers a wide range of applications: reflux condensations,reactions under vacuum, crystallizations, (pH-controlled) precipitation reactions, nanoparticle
or catalyst syntheses, and the transfer from laboratory to process scale.

Individual application support
In the IKA Application Center, you can see the capabilities ofour reactor systems for yourself. Our experts will show you
the application and demonstrate the extended accessories.

One service for everything,worldwideYour decision for EasySyn from IKA is also a decision for a single
contact partner for all reactor components. All components, fromthe overhead stirrers to the vessels to the temperature control
system, are made by us. This means you always benefit fromIKA's excellent technical service in your region. IKA guarantees
the availability of spare parts for your laboratory device for up to10 years.

Do you have any questions?
Our service team is at your personal disposal:

00 8000 4524357 (00 8000 IKAHELP)
2+1 years
warranty** 2+1 years after registering at
www.ika.com/register,wearing parts and glassware excluded.

EasySyn | 3

Everything from a single source/// One reactor system, all from IKA
EasySyn is the reactor system for organic or aqueous synthesis.
And the only reactor system for which you can get all thecomplementary components from a single source: Stirrers,circulators, vacuum pumps and controllers, sensors andautomation software. All from IKA and all perfectly matched toeach other.
With its double-walled reactor vessels from 500 ml to 5000 ml,
EasySyn is a medium-sized and very flexible upscaling system. It
ensures safe and reproducible chemical reactions.

411/// OVERVIEW/// ACCESSORIES

614/// KEY COMPONENTS/// PACKAGES

1015/// MORE FEATURES/// TECHNICAL DATA

4 | Overview

For the best reactions
One stand for all vessels
EasySyn has a stand that fits all vessel sizes. It is ergonomicallydesigned. This means that the reactor vessels can be adjusted
in height without tools. Draining the reaction liquid is easy,changing vessels is quick and the stirrer can be aligned in threedimensions.
This stand supports you through the entire upscaling process.
You have a comfortable working posture at all times, and thevarious operations go off the hook more quickly.

Easy to use
EasySyn ensures safe handling of the stirring tool at all times. Our
specially developed vessel clamps enable tool-free and simpleattachment of the vessels.
You can assemble and reassemble EasySyn very quickly. In
addition, the straight design guarantees a long service life.

Thought-out reactor vessels
These reactor vessels have what it takes. They are available with
500, 1000, 2000 and 5000 milliliter volumes and with or without
a bottom drain valve. The spring-loaded valve spindle preventsstresses in the glass and thus glass breakage. All vessels are
made of borosilicate glass 3.3, which means they are chemicallyresistant.

500 – 5000 milliliter volume

EasySyn allows you to run completely different batch sizes. You
work securely and save costs. The bottom valve ensures complete
draining of the reaction solution and avoids dead volumes.

Overview | 5

Powerful stirrers
Suitable for EasySyn, our overhead stirrers are available withtorques of 60 or 100 Ncm. You can choose from various stirring
tools adapted to the reactor vessel, all of which are PTFE-coated.
And if you choose a control unit, you can remotely control thestirring parameters thanks to the removable Wireless Controller
and use many additional functions.
Your application will certainly succeed thanks to powerful mixing.
You can automate numerous work steps and, for example,
control them remotely from outside a fume hood.

-60 – 230 °C

30 000 mPas3 mbar

Large operating range
EasySyn is capable of operating at temperatures between -60 °C
and 230 °C. The reactor system handles liquids up to about
30 000 mPas and vacuum applications up to 3 mbar.
EasySyn offers you a wide range of applications. It is optimized
for media with low viscosity, but is very flexible in everyday use.

Complete compatibility
Our reactor lids are normed. The additional openings for sensors
or glass devices are also compatible with available equipment.
EasySyn is flexible and compatible with your existing equipment.

6 | Key components

Key components
Overhead stirrers
In combination with EasySyn, you use our proven EUROSTAR overhead stirrers.
You have the choice between three device classes and strengths: There is ourclassic EUROSTAR 60 digital stirrer, the fully automatable EUROSTAR 60 control
or the extra strong EUROSTAR 100 control.
All three device variants are universal and robust overhead stirrers. In the 100control variant, EUROSTAR achieves an extremely high torque of up to 100 Ncm.
This makes it particularly popular for applications in laboratory reactors.

Overhead stirrers

EUROSTAR 60 digital
Ident No.: 0004446000

EUROSTAR 60 control
Ident No.: 0004440000

EUROSTAR 100 control
Ident No.: 0004028500

Torque max. at the stirring shaft

60 Ncm60 Ncm100 Ncm

RS 232 / USB interface

–

Yes

Yes

Removable Wireless Controller

–

Yes

Yes

Programmable functions

–

Yes

Yes

Reversible direction of rotation

––

Yes

Key components | 7

Stirring tools
You can clamp our PTFE-coated propeller stirrers or anchor stirrers into theoverhead stirrer of your choice. They are available in different sizes – matching
the chosen glass vessel.

> STIRRING ELEMENTS, AXIAL FLOW

Propeller stirrer

R 4011 SY
Ident No.: 0020114111

R 4012 SY
Ident No.: 0020114112

R 4013 SY
Ident No.: 0020114113

Stirrer diameter

40 mm

50 mm

75 mm

Shaft length

400 mm

480 mm

530 mmSuitable vessel size

500 / 1000 ml

1000 / 2000 ml

5000 ml> STIRRING ELEMENTS, TANGENTIAL FLOWAnchor stirrer

R 4021 SY
Ident No.: 0020114114

R 4022 SYIdent No.: 0020114115

Stirrer diameter

89 mm

136 mmShaft length

410 mm

530 mmSuitable vessel size

500 / 1000 ml

2000 / 5000 ml

8 | Key components

Reactor vessels
In total, you have eight different reactor vessels to choose from. There are four
volumes: 500, 1000, 2000 and 5000 ml. Each size is optionally available with orwithout a bottom drain valve. As the vessels are made of borosilicate glass 3.3,
EasySyn is also suitable for syntheses with aggressive media such as acids or bases.
If you decide to use the drain valve, simply release the complete reaction solution
from the bottom without leaving any residue. This makes it easier to clean thevessels later. The spring loaded valve spindle is standard on all our reactor vessels
to prevent stresses in the glass and glass breakage.

> REACTOR VESSELS

500
milliliter
volume1000
milliliter
volume2000
milliliter
volume5000
milliliter
volumeReactor vessels

SY 500 | SY 500 D
Ident No.: 0020113459
Ident No.: 0020113460

SY 1000 | SY 1000 D
Ident No.: 0020113461
Ident No.: 0020113462

SY 2000 | SY 2000 D
Ident No.: 0020113463
Ident No.: 0020113464

SY 5000 | SY 5000 D
Ident No.: 0020113465
Ident No.: 0020113466

Volume500 ml

1000 ml2000 ml

5000 ml

Bottom drain valve

no | yesno | yesno | yesno | yes

Flange size

DN 100

DN 100

DN 150

DN 150Min. operating temperature-60 °C | -50 °C-60 °C | -50 °C-60 °C | -50 °C-60 °C | -50 °C

Max. operating temperature

230 °C | 200 °C

230 °C | 200 °C

230 °C | 200 °C

230 °C | 200 °C

Key components | 9Reactor lids
You can select from three different reactor lids. The stirring tool is inserted inhere and temperature or pH sensors can be connected easily. The shape andground joints of the reactor lids are standardized and accordingly compatible
with everything you already have.

> REACTOR LIDS

SY 100.1

SY 150.1

SY 150.2

SY 100.1
Ident No.: 0020113649

SY 150.1
Ident No.: 0020113660

SY 150.2
Ident No.: 0020113661

Openings

1 × center NS 29/32,
2 × NS 29/32,
1 × NS 14/23

1 × center NS 29/32,
2 × NS 29/32,
1 × NS 14/23

1 × center NS 29/32,3 × NS 29/32,
1 × NS 14/23, 2 × GL 18

Flange size

DN 100

DN 150

DN 150Suitable vessel size

500 / 1000 ml

2000 / 5000 ml

2000 / 5000 ml

Reactor lids

10 | More features

Even more features
Flexible tempering hoses
Our tempering hoses are particularly flexible. This means they
ideally nestle around the reactor without tension. This avoids
stresses on the glass vessel and thus glass breakage. You will
work safer and more cost-effectively.

Clever valves
In order to drain the temperature fluid completely and cleanly
from the double-walled vessel and the tempering hoses, we havedeveloped special valves that are attached to the stand.

Stirrer in three dimensions
Thanks to the stand design, our stirrer can be moved in threedimensions. This means that the reactor lid is always easily
accessible. You can fill or remove your substances effortlessly.
This safe alignment also ensures a long service life of the stirrer
guide.

Easy cleaning
A clear design and completely emptied vessels relieve yourstaff of time-consuming cleaning tasks in everyday laboratory
work and at the same time ensure safe working.

Accessories | 11

Perfect complements
Automation with labworldsoft® 6Automate, control, monitor: With our labworldsoft® 6 laboratory software, you operate your entire
reactor system from your PC. You define dependencies, automate routine tasks and define measured value recordings or real-time calculations.
In remote operation, you control sophisticated reactions from a safe distance. Several options forrecipe and process automation ensure greater process reliability, more user safety and higher reproducibility of all processes.

labworldsoft® 6 Starterlabworldsoft® 6 Advancedlabworldsoft® 6 Pro

Ident No.: 0020019397

Ident No.: 0020105873

Ident No.: 0020017366› For small device
configurations› For up to 3 devices› For laboratory devices of all
manufacturers› 1 year support with newversions oflabworldsoft® 6 Starter› For all device configurations› More than 700 devices fromdevice library› Control any number of devices simultaneously› For laboratory devices of all
manufacturers› 1 year support with newversions oflabworldsoft® 6 Advanced› For all device configurations› More than 700 devices fromdevice library› Control any number of devices simultaneously› For laboratory devices of allmanufacturers› 1 year support with new versions of labworldsoft® 6 Pro› Supports the Labworldsoft
server› Supports FDA 21 CFR Part 11requirements

12 | Accessories

Temperature control

Circulators

HBC 5 control
Ident No.: 0004127000

HRC 2 control
Ident No.: 0025004524

CBC 5 control
Ident No.: 0004167000

Operating temperature

RT +10 K – 250 °C-30 – 100 °C-25 – 200 °C

Application

Heating / cooling(with external cooling)

Heating / cooling

Heating / cooling

Heating capacity

2500 W

1500 W

2500 W

Cooling capacity (@20 °C)–400 W

350 W

Vacuum pump

MVP 10 basic
Ident No.: 0003980000

VACSTAR digital
Ident No.: 0020016236

VACSTAR control
Ident No.: 0020109375

Max. pump rate

28.3 l/min

22 l/min22 l/minFinal pressure

7 mbar

2 mbar

2 mbarSpeed control

ON / OFF

0/285 – 1200 rpm

0/285 – 1200 rpm

Adjustable vacuum value

Requires external
control / valve

Requires external
control / valve

Integrated controller

Vacuum pumps

VC 10 lite | VC 10 pro
Ident No.: 0020111998 | 0020112031

Universal vacuum controller with on/off regulation for
calibration of multiple vacuum sources. In addition tothe manual mode, up to ten programs can be saved,with up to ten pressure/time sequences.

Cleaning functionof valves for longservice lifeOther accessories | 13

Other accessories

PT 100.300
PTFE-coated temperature sensor.
Length: 300 mm
Suitable for reactor vessels withvolumes of 500 ml and 1000 ml.
Ident No.: 0020114304

PT 100.500
PTFE-coated temperature sensor.
Length: 500 mm
Suitable for reactor vessels withvolumes of 2000 ml and 5000 ml.
Ident No.: 0020114305

SY pH 325
pH electrode
Length: 325 mm
Suitable for reactor vessels withvolumes of 500 ml and 1000 ml.
Ident No.: 0020114414

SY pH 425
pH electrode
Length: 425 mm
Suitable for reactor vessels withvolumes of 2000 ml and 5000 ml.
Ident No.: 0020114415

LT 5.40
Draining set
By using the draining set, easy andclean draining of the temperature
fluid is made possible.
Ident No.: 0020113266

SY NS.6 | 8 | 12
Swivel screw fitting
Diameter: 6 mm | 8 mm | 12 mm
Adapter to mount temperature
sensors (SY NS.8) or pH electrodes(SY NS.12).
Ident No.: 0020113165
Ident No.: 0020113164
Ident No.: 0020113163

LT 5.30 | 31 | 32
Tempering hose, 2 pcs.
Length: 800 | 1200 | 1600 mm
Temperature range: -60 – 300 °C
Pressure resistance: 3 bar
Ident No.: 0020108012
Ident No.: 0020110278
Ident No.: 0020113271

H.FKM.12
Hose set DN 12, 2 pcs.
Length: 1500 mm
Temperature range: -30 – 180 °C
Pressure resistance: 1 bar
Ident No.: 0004569300

Further accessories
can be found at
www.ika.com

SY C1
Reflux condenser with NS 29/32 core.
Ø Olives: 11 mm
Ident No.: 0020113927

ALMEMO 2490
pH measuring device
Ident No.: 0004709300

14 | Packages

Your laboratory, our package
We have prepared a Starter and an Advanced Package for you. The main difference lies in the stirrer
used. EasySyn Starter contains the EUROSTAR 60 digital, the Advanced Packages the EUROSTAR 100
control. It enables additional functions such as external control via labworldsoft® 6 (RS 232 or USB
interface), operation via detachable remote control and programming of a timer, for example.
The Advanced Systems also include the LT 5.40 draining set, which enables the temperature fluid of
the double jacket to be drained easily and cleanly by using special valves. This saves valuable time
when replacing the reactor vessels.
The Advanced System is therefore particularly suitable for applications in which the reactor vessel
has to be replaced or cleaned regularly.

EasySyn Starter

EasySyn Advanced
Stand system

Double-walled reaction vessel with bottom outlet.
(Sizes: 500, 1000, 2000 or 5000 ml)
Reactor lid
Propeller stirrer incl. stirrer guide
Required clamps for vessel mounting & lid attachment
Overhead Stirrer EUROSTAR 60 digital

Overhead Stirrer EUROSTAR 100 control
Draining set LT 5.40

1 EasySyn 500 Starter

Ident No.: 0020113433

EasySyn 500 AdvancedIdent No.: 0020113434

EasySyn 1000 Starter Ident No.: 0020113435

2 EasySyn 1000 Advanced Ident No.: 0020113436

3 EasySyn 2000 Starter Ident No.: 0020113438

EasySyn 2000 Advanced Ident No.: 0020113440

EasySyn 5000 Starter Ident No.: 0020113442

4 EasySyn 5000 Advanced Ident No.: 0020113443

1324

Technical data | 15

Technical data
TECHNICAL DATA

EasySyn Starter

EasySyn AdvancedUseful volume max.

500 ml, 1000 ml, 2000 ml, 5000 ml

Operating temperature-50 – 200 °C

Attainable vacuum
Openings reactor vessel

3 mbar1 × center NS 29/32, 2 × NS 29/32, 1 × NS 14/23

Stand dimensions (W × H × D)

500 × 950 × 580 mm

Draining set LT 5.40–

inclusive

EUROSTAR 60 digital

EUROSTAR 100 control

60 Ncm100 Ncm

Timer

–

Yes

RS 232 interface–

Yes

USB interface–

Yes

Reversible direction of rotation–

Yes

Torque display–

Yes

Overhead stirrer
Torque max. at the stirrer shaft

Would you like your own desired configuration?
Or an individual offer?
With our online configurator, you can put together a reactor
system that is individually tailored to your needs.
Click here to view the configurator:

www.ika.com/easysyn-config
You can choose between different reactor vessels, overhead stirrers,
stirring tools, circulators and automation levels.

EasySyn configurator

ENGERMANY
IKA-Werke GmbH & Co. KG
Janke & Kunkel-Straße 10, 79219 Staufen
Phone: +49 7633 831-0, eMail: sales@ika.deUSA
IKA Works, Inc.
Phone: +1 910 452-7059
eMail: sales@ika.net

KOREA
IKA Korea Ltd.
Phone: +82 2 2136 6800
eMail: info@ika.kr

BRAZILIKA Brazil
Phone: +55 19 3772-9600
eMail: sales@ika.net.br

MALAYSIA
IKA Works (Asia) Sdn Bhd
Phone: +60 3 6099-5666
eMail: sales.lab@ika.my

CHINA
IKA Works Guangzhou
Phone: +86 20 8222 6771
eMail: info@ika.cn

POLANDIKA Poland Sp. z o.o.
Phone: +48 22 201 99 79
eMail: sales.poland@ika.com

JAPANIKA Japan K.K.
Phone: +81 6 6730 6781
eMail: info_japan@ika.ne.jp

INDIA
IKA India Private Limited
Phone: +91 80 26253 900
eMail: info@ika.in

UNITED KINGDOMIKA England LTD.
Phone: +44 1865 986 162
eMail: sales.england@ika.com

VIETNAMIKA Vietnam Company Limited
Phone: +84 28 38202142
eMail: sales.lab-vietnam@ika.com

THAILANDIKA Works (Thailand) Co. Ltd.
Phone: +66 86 375 7451
eMail: sales.lab-thailand@ika.com

TURKEYIKA Turkey
Phone: +90 216 394 43 43
eMail: sales.turkey@ika.com

Subject to technical changes.
Information regarding delivery is not binding.

www.ika.com

IKAworldwide // #lookattheblue

202208_EasySyn_Brochure_IWS_EN/// WORLDWIDE