6K-ERB08

Specifications

Document Revision 1.1, February, 2010
© Copyright 2010, Measurement Computing Corporation

Specifications
Typical for 25 °C unless otherwise specified.
Specifications in italic text are guaranteed by design.

Power consumption
Table 1. Power consumption specifications
5 V PC auxiliary power / PCI bus power /external regulated
External 7.5 V to 15 V unregulated supply

All relays off
All relays on
All relays off
All relays on

15 mA typical, 20 mA max.
480 mA typical, 550 mA max.
20 mA typical, 25 mA max.
500 mA typical, 580 mA max.

Output specification
Table 2. Output specifications
NumberContact configuration

88 Form C (SPDT) NO, NC and Common available at screw terminals

Contact rating
Contact resistanceOperate time
Release time
Vibration
Shock
Dielectric isolation (between open contact)
Life expectancy
Power on state (no connection to DIO)

6 A @ 120 VAC or 28 VDC resistive
100 milliohms max (initial value)
10 milliseconds max
5 milliseconds max
10 to 55 Hz (amplitude 1.5 mm)
10 G (11 milliseconds)
300 VAC, 50/60 Hz (1 minute)10 million mechanical operations, min
Not energized. NC in contact to Common. (JP2 in NON-INVERT position)

Environmental
Table 3. Environmental specifications
Operating temperature range
Storage temperature range
Humidity

0 to 70 °C-40 to 100 °C
0 to 95% non-condensing

Mechanical
Table 4. Mechanical specifications
Card dimensions (without standoff)
Weight229 mm (L) x 102 mm (W) x 20 mm (H)
9" (L) x 4" (W) x 0.8" (H)
0.6 lb.

1

6K-ERB08

Specifications

Relay screw terminals
Table 5. Relay screw terminal specifications
Wire gauge range

12 AWG to 20 AWG

Table 6. Screw terminal pin outPinSignal Name

1-NO

Relay 1 Normally Open contact

1-C

Relay 1 Common contact

1-NC

Relay 1 Normally Closed contact

2-NO

Relay 2 Normally Open contact

2-C

Relay 2 Common contact

2-NC

Relay 2 Normally Closed contact

3-NO

Relay 3 Normally Open contact

3-C

Relay 3 Common contact

3-NC

Relay 3 Normally Closed contact

4-NO

Relay 4 Normally Open contact

4-C

Relay 4 Common contact

4-NC

Relay 4 Normally Closed contact

5-NO

Relay 5 Normally Open contact

5-C

Relay 5 Common contact

5-NC

Relay 5 Normally Closed contact

6-NO

Relay 6 Normally Open contact

6-C

Relay 6 Common contact

6-NC

Relay 6 Normally Closed contact

7-NO

Relay 7 Normally Open contact

7-C

Relay 7 Common contact

7-NC

Relay 7 Normally Closed contact

8-NO

Relay 8 Normally Open contact

8-C

Relay 8 Common contact

8-NC

Relay 8 Normally Closed contact

Relay logic jumper (JP2)
Table 7. JP2 specifications
Invert (1-2)
Non-invert (2-3) (default)

Relay activates when DIO is LOW
Relay activates when DIO is HIGH

Power-in jumper (JP1)
Table 8. JP1 specifications+5PC+9V EXT+5BD (default)

Use cable C-PCPOWER-10
Use Adapter CB-PWR-9
Powered from 100-pin connector

2

6K-ERB08

Specifications

Relays pull-up/down option
Table 9. Relay pull-up/down specifications
R21,23,25,27,29,31,33,35
R20,22,24,26,28,30,32,34

Relays NC pin pull-up/down
Relays NO pin pull-up/downCompatible products
Table 10. Compatible product specifications
Analog input boards

Analog output boards

 PCI-DAS6013
 PCI-DAS6014
 PCI-DAS6030
 PCI-DAS6031
 PCI-DAS6032
 PCI-DAS6033
 PCI-DAS6034
 PCI-DAS6035
 PCI-DAS6036
 PCI-DAS6052
 PCI-DAS6023
 PCI-DAS6025
 PCI-DAS6040
 PCI-DAS6070
 PCI-DAS6071
 PCI-DAC6702
 PCI-DAC6703

Note 1: The 6K-ERB08 requires external power (for all products above) when used with C100HD50 (pins51-100) ribbon cable.

3

6K-ERB08

Specifications

Main connectors and pin out
User connector P20
Table 11. Connector P20 specifications
Connector typeCompatible cables

Shielded SCSI 100 D-type
C100MMS-x, shielded round cable. x = 1, 2 or 3 meters
Table 12. P20 pin outPin1234567891011121314151617181920212223242526272829303132333435363738394041424344454647484950Signal name
P21 Pass Through 1
P21 Pass Through 2
P21 Pass Through 3
P21 Pass Through 4
P21 Pass Through 5
P21 Pass Through 6
P21 Pass Through 7
P21 Pass Through 8
P21 Pass Through 9
P21 Pass Through 10
P21 Pass Through 11
P21 Pass Through 12
P21 Pass Through 13
P21 Pass Through 14
P21 Pass Through 15
P21 Pass Through 16
P21 Pass Through 17
P21 Pass Through 18
P21 Pass Through 19
P21 Pass Through 20
P21 Pass Through 21
P21 Pass Through 22P21 Pass Through 23
P21 Pass Through 24
P21 Pass Through 25
P21 Pass Through 26
P21 Pass Through 27
P21 Pass Through 28
P21 Pass Through 29
P21 Pass Through 30
P21 Pass Through 31
P21 Pass Through 32
P21 Pass Through 33
P21 Pass Through 34
P21 Pass Through 35
P21 Pass Through 36
P21 Pass Through 37
P21 Pass Through 38
PC +5VP21 Pass Through 40
P21 Pass Through 41
P21 Pass Through 42
P21 Pass Through 43
P21 Pass Through 44
P21 Pass Through 45
P21 Pass Through 46
P21 Pass Through 47
P21 Pass Through 48
P21 Pass Through 49GNDPin51
52
53
54
55
56
57
58
59
60
6162
63
64
65
66
67
68
69707172737475
76
7778
79
80
81828384858687
88899091
9293949596
979899100

4Signal name
P21 Pass Through 51
P21 Pass Through 52
P21 Pass Through 53
P21 Pass Through 54
P21 Pass Through 55
P21 Pass Through 56
P21 Pass Through 57
P21 Pass Through 58
P21 Pass Through 59
P21 Pass Through 60
P21 Pass Through 61
P21 Pass Through 62
P21 Pass Through 63
P21 Pass Through 64
P21 Pass Through 65
P21 Pass Through 66
P21 Pass Through 67P21 Pass Through 68
P21 Pass Through 69
P21 Pass Through 70
P21 Pass Through 71
P21 Pass Through 72
P21 Pass Through 73
P21 Pass Through 74
P21 Pass Through 75
P21 Pass Through 76
P21 Pass Through 77
P21 Pass Through 78
P21 Pass Through 79
P21 Pass Through 80
P21 Pass Through 81
P21 Pass Through 82
P21 Pass Through 83
P21 Pass Through 84
DIO0
DIO1
DIO2
DIO3
DIO4
DIO5
DIO6
DIO7
P21 Pass Through 93
P21 Pass Through 94
P21 Pass Through 95
P21 Pass Through 96
P21 Pass Through 97
P21 Pass Through 98
P21 Pass Through 99GND

6K-ERB08

Specifications

User connector P21
Table 13. Connector P21 specifications
Connector typeCompatible cables

Shielded SCSI 100 D-type
C100MMS-x, shielded round cable. x =1, 2 or 3 meters
Table 14. P21 pin outPin1234
5678
91011121314
15
16
1718
192021
22232425262728
2930
3132
33
34
35
36
37
38
39
40
41
42
43
4445
46
47
48
4950Signal name
P20 Pass Through 1
P20 Pass Through 2
P20 Pass Through 3
P20 Pass Through 4
P20 Pass Through 5
P20 Pass Through 6
P20 Pass Through 7
P20 Pass Through 8
P20 Pass Through 9
P20 Pass Through 10
P20 Pass Through 11
P20 Pass Through 12
P20 Pass Through 13
P20 Pass Through 14
P20 Pass Through 15
P20 Pass Through 16
P20 Pass Through 17
P20 Pass Through 18
P20 Pass Through 19
P20 Pass Through 20
P20 Pass Through 21
P20 Pass Through 22
P20 Pass Through 23
P20 Pass Through 24
P20 Pass Through 25
P20 Pass Through 26
P20 Pass Through 27
P20 Pass Through 28
P20 Pass Through 29
P20 Pass Through 30
P20 Pass Through 31
P20 Pass Through 32
P20 Pass Through 33
P20 Pass Through 34
P20 Pass Through 35
P20 Pass Through 36
P20 Pass Through 37
P20 Pass Through 38
PC +5VP20 Pass Through 40
P20 Pass Through 41
P20 Pass Through 42
P20 Pass Through 43
P20 Pass Through 44
P20 Pass Through 45
P20 Pass Through 46
P20 Pass Through 47
P20 Pass Through 48
P20 Pass Through 49
GNDPin
51
52
53
54
55
56
57
58
59
60
61
62
63
64
65
66
67
68
69
70
71
72
73
74
75
76
77
78
79
80
81
82
83
84
85
86
87
88
89
90
91
92
939495
96
97
98
99
100

5Signal name
P20 Pass Through 51
P20 Pass Through 52
P20 Pass Through 53
P20 Pass Through 54
P20 Pass Through 55
P20 Pass Through 56
P20 Pass Through 57
P20 Pass Through 58
P20 Pass Through 59
P20 Pass Through 60
P20 Pass Through 61
P20 Pass Through 62
P20 Pass Through 63
P20 Pass Through 64
P20 Pass Through 65
P20 Pass Through 66
P20 Pass Through 67
P20 Pass Through 68
P20 Pass Through 69
P20 Pass Through 70
P20 Pass Through 71
P20 Pass Through 72
P20 Pass Through 73
P20 Pass Through 74
P20 Pass Through 75
P20 Pass Through 76
P20 Pass Through 77
P20 Pass Through 78
P20 Pass Through 79
P20 Pass Through 80
P20 Pass Through 81
P20 Pass Through 82
P20 Pass Through 83
P20 Pass Through 84
DIO0
DIO1
DIO2
DIO3
DIO4
DIO5
DIO6
DIO7
P20 Pass Through 93
P20 Pass Through 94
P20 Pass Through 95
P20 Pass Through 96
P20 Pass Through 97
P20 Pass Through 98
P20 Pass Through 99
GND

6K-ERB08

Specifications

User connector P5
Table 15. Connector P5 specifications
Connector typeCompatible cables

Unshielded 50-pin ribbon connector - male
C100HD50-x, C50FF-x, unshielded ribbon cable. x = 3 or 6 feet
Table 16. P5 pin outPin12345678910111213141516171819202122232425Signal name
P6 Pass Through 51
P6 Pass Through 52
P6 Pass Through 53
P6 Pass Through 54
P6 Pass Through 55
P6 Pass Through 56
P6 Pass Through 57
P6 Pass Through 58
P6 Pass Through 59
P6 Pass Through 60
P6 Pass Through 61
P6 Pass Through 62
P6 Pass Through 63
P6 Pass Through 64
P6 Pass Through 65
P6 Pass Through 66
P6 Pass Through 67
P6 Pass Through 68
P6 Pass Through 69
P6 Pass Through 70
P6 Pass Through 71
P6 Pass Through 72
P6 Pass Through 73
P6 Pass Through 74
P6 Pass Through 75

Pin26272829303132333435363738394041424344454647484950

6Signal name
P6 Pass Through 76
P6 Pass Through 77
P6 Pass Through 78
P6 Pass Through 79
P6 Pass Through 80
P6 Pass Through 81
P6 Pass Through 82
P6 Pass Through 83
P6 Pass Through 84
DIO0
DIO1
DIO2
DIO3
DIO4
DIO5
DIO6
DIO7
P6 Pass Through 93
P6 Pass Through 94
P6 Pass Through 95
P6 Pass Through 96
P6 Pass Through 97
P6 Pass Through 98
P6 Pass Through 99
GND

6K-ERB08

Specifications

User connector P6Table 17. Connector P6 specifications
Connector typeCompatible cables

Unshielded 50 pin ribbon connector - male
C100HD50-x, C50FF-x, unshielded ribbon cable. x = 3 or 6 feet
Table 18. P6 pin outPin
123456789101112131415161718
19202122232425Signal name
P5 Pass Through 51
P5 Pass Through 52
P5 Pass Through 53
P5 Pass Through 54
P5 Pass Through 55
P5 Pass Through 56
P5 Pass Through 57
P5 Pass Through 58
P5 Pass Through 59
P5 Pass Through 60
P5 Pass Through 61
P5 Pass Through 62
P5 Pass Through 63
P5 Pass Through 64
P5 Pass Through 65
P5 Pass Through 66
P5 Pass Through 67
P5 Pass Through 68
P5 Pass Through 69
P5 Pass Through 70
P5 Pass Through 71
P5 Pass Through 72
P5 Pass Through 73
P5 Pass Through 74
P5 Pass Through 75

Pin
26
2728
2930
3132
33
34
35
36
37
38
3940
41
42
43
4445
46
47
48
49
50

7Signal name
P5 Pass Through 76
P5 Pass Through 77
P5 Pass Through 78
P5 Pass Through 79
P5 Pass Through 80
P5 Pass Through 81
P5 Pass Through 82
P5 Pass Through 83
P5 Pass Through 84
DIO0
DIO1
DIO2
DIO3
DIO4
DIO5
DIO6
DIO7
P5 Pass Through 93
P5 Pass Through 94
P5 Pass Through 95
P5 Pass Through 96
P5 Pass Through 97
P5 Pass Through 98
P5 Pass Through 99
GND

Measurement Computing Corporation
10 Commerce Way
Suite 1008
Norton, Massachusetts 02766(508) 946-5100
Fax: (508) 946-9500
E-mail: info@mccdaq.com
www.mccdaq.com
6K-ERB08-spec.doc