PCI-DAS1002
Multifunction Analog & Digital I/O
User's Guide

Document Revision 5A, March, 2009
© Copyright 2009, Measurement Computing Corporation

Trademark and Copyright Information
Measurement Computing Corporation, InstaCal, Universal Library, and the Measurement Computing logo areeither trademarks or registered trademarks of Measurement Computing Corporation. Refer to the Copyrights &Trademarks section on mccdaq.com/legal for more information about Measurement Computing trademarks.
Other product and company names mentioned herein are trademarks or trade names of their respective
companies.
© 2000 Measurement Computing Corporation. All rights reserved. No part of this publication may be
reproduced, stored in a retrieval system, or transmitted, in any form by any means, electronic, mechanical, by
photocopying, recording, or otherwise without the prior written permission of Measurement Computing
Corporation.
NoticeMeasurement Computing Corporation does not authorize any Measurement Computing Corporation product for
use in life support systems and/or devices without prior written consent from Measurement Computing
Corporation. Life support devices/systems are devices or systems that, a) are intended for surgical implantation
into the body, or b) support or sustain life and whose failure to perform can be reasonably expected to result ininjury. Measurement Computing Corporation products are not designed with the components required, and are
not subject to the testing required to ensure a level of reliability suitable for the treatment and diagnosis ofpeople.

HM PCI-DAS1002.doc

3

4

Table of Contents
Preface

About this User's Guide .......................................................................................................................7
What you will learn from this user's guide ......................................................................................................... 7
Conventions in this user's guide ......................................................................................................................... 7
Where to find more information ......................................................................................................................... 7
Register-level programming ............................................................................................................................... 8
Chapter 1

Introducing the PCI-DAS1002 ..............................................................................................................9
Overview: PCI-DAS1002 features ..................................................................................................................... 9
Software features ................................................................................................................................................ 9
PCI-DAS1002 block diagram ........................................................................................................................... 10
Chapter 2

Installing the PCI-DAS1002 ................................................................................................................11
What is included with your PCI-DAS1002 ...................................................................................................... 11
Hardware .........................................................................................................................................................................11
Additional documentation ................................................................................................................................................11
Optional components .......................................................................................................................................................11

Unpacking the PCI-DAS1002 .......................................................................................................................... 12
Installing the software ...................................................................................................................................... 12
Installing the hardware ..................................................................................................................................... 12
Configuring the hardware ................................................................................................................................. 13
Differential input mode ....................................................................................................................................................13
Single-ended input mode .................................................................................................................................................13

Connecting the board for I/O operations .......................................................................................................... 13
Connectors, cables – main I/O connector.........................................................................................................................13
Pinout – main I/O connector ............................................................................................................................................14
Cables ..............................................................................................................................................................................16
Field wiring and signal termination .................................................................................................................................16
Chapter 3

Calibrating the Board..........................................................................................................................17
Introduction ...................................................................................................................................................... 17
Calibration theory ............................................................................................................................................. 17
Chapter 4

Specifications ......................................................................................................................................18
Analog input ..................................................................................................................................................... 18
Accuracy ..........................................................................................................................................................................19
Crosstalk ..........................................................................................................................................................................19
Noise performance ...........................................................................................................................................................19

Analog output ................................................................................................................................................... 20
Absolute accuracy ............................................................................................................................................................20
Calibrated accuracy components......................................................................................................................................20

Counters............................................................................................................................................................ 21
Digital input/output........................................................................................................................................... 22
Interrupts........................................................................................................................................................... 22
Miscellaneous ................................................................................................................................................... 22Power consumption .......................................................................................................................................... 22
Environmental .................................................................................................................................................. 22
Mechanical ....................................................................................................................................................... 225

PCI-DAS1002 User's Manual

Connector and pin out....................................................................................................................................... 23
8-channel differential mode pin out .................................................................................................................................23
16-channel single-ended mode pin out.............................................................................................................................24

Declaration of Conformity ..................................................................................................................25

6

Preface

About this User's Guide
What you will learn from this user's guideThis user's guide explains how to install, configure, and use the PCI-DAS1002 so that you get the most out of
its analog, digital, and timing I/O features.
This user's guide also refers you to related documents available on our web site, and to technical support
resources.

Conventions in this user's guide
For more information on …
Text presented in a box signifies additional information and helpful hints related to the subject matter you arereading.
Caution! Shaded caution statements present information to help you avoid injuring yourself and others,
damaging your hardware, or losing your data.
< : >Angle brackets that enclose numbers separated by a colon signify a range of numbers, such as those assigned
to registers, bit settings, etc.

bold text

Bold text is used for the names of objects on the screen, such as buttons, text boxes, and check boxes. For
example:1. Insert the disk or CD and click the OK button.

italic text

Italic text is used for the names of manuals and help topic titles, and to emphasize a word or phrase. For
example:The InstaCal® installation procedure is explained in the Quick Start Guide.
Never touch the exposed pins or circuit connections on the board.

Where to find more information
The following electronic documents provide relevant information to the operation of your PCI-DAS1002.







MCC's Specifications: PCI-DAS1002 (the PDF version of the Specifications chapter in this guide) isavailable on our web site at www.mccdaq.com/pdfs/Specs/PCI-DAS1002-spec.pdf.
MCC's Quick Start Guide is available on our web site atwww.mccdaq.com/PDFmanuals/DAQ-Software-Quick-Start.pdf.
MCC's Guide to Signal Connections is available on our web site atwww.mccdaq.com/signals/signals.pdf.
MCC's Universal Library User's Guide is available on our web site atwww.mccdaq.com/PDFmanuals/sm-ul-user-guide.pdf.
MCC's Universal Library Function Reference is available on our web site atwww.mccdaq.com/PDFmanuals/sm-ul-functions.pdf.
MCC's Universal Library for LabVIEW™ User’s Guide is available on our web site atwww.mccdaq.com/PDFmanuals/SM-UL-LabVIEW.pdf.

PCI-DAS1002 User's Manual (this document) is also available on our web site atwww.mccdaq.com/PDFmanuals/PCI-DAS1002.pdf.

7

PCI-DAS1002 User's Manual

About this User's Guide

Register-level programming
You should use the Universal Library to control your board. Only experienced programmers should perform
register-level programming.
If you need to program at the register level in your application, refer to the Register Map for the PCI-DAS1000
Series (available at www.mccdaq.com/registermaps/RegMapSTC1000.pdf).

8Chapter 1

Introducing the PCI-DAS1002
Overview: PCI-DAS1002 features
This manual explains how to install and use the PCI-DAS1002. This board is a multifunction analog and digital
I/O board designed for the PCI bus.
The PCI-DAS1002 provides either 16 single-ended or eight differential analog inputs. Input ranges aresoftware-selectable as either bipolar or unipolar.
Analog input ranges for the PCI-DAS1002 are software programmable for the following ranges:


Bipolar: ±10 V, ±5 V, ±2.5 V, and ±1.25 V
Unipolar: 0 to 10 V, 0 to 5 V, 0 to 2.5 V and 0 to 1.25 V

The PCI-DAS1002 provides a 200 kHz sample rate.
The PCI-DAS1002 also provides two channels of 12-bit analog output. Analog output ranges are ±10 V and±5 V (bipolar) and 0-10 V, 0-5 V (unipolar).
The PCI-DAS1002 offers 24-bits of parallel, digital I/O (two 8-bit ports and two 4-bit ports). You can configure
each port independently as input or output.
This board has an 82C54 counter chip which has three user-accessible 16-bit counters. Clock, gate, and output
signals from two of three counters are available on a 100-pin connector. The third counter is used internally.
The PCI-DAS1002 is fully plug-and-play. All addresses and interrupt channels are set with software.

Software features
For information on the features of InstaCal and the other software included with your PCI-DAS1002, refer tothe Quick Start Guide that shipped with your device. The Quick Start Guide is also available in PDF at
www.mccdaq.com/PDFmanuals/DAQ-Software-Quick-Start.pdf.
Check www.mccdaq.com/download.htm for the latest software version.

9

PCI-DAS1002 User's Manual

Introducing the PCI-DAS1002

PCI-DAS1002 block diagram
PCI-DAS1002 functions are illustrated in the block diagram shown here.

Figure 1. PCI-DAS1002 functional block diagram

10Chapter 2

Installing the PCI-DAS1002
What is included with your PCI-DAS1002
As you unpack your board, make sure each of the items shown below is included:

Hardware

PCI-DAS1002

Additional documentation
In addition to this hardware user's guide, you should also receive the Quick Start Guide (available in PDF at
www.mccdaq.com/PDFmanuals/DAQ-Software-Quick-Start.pdf). This booklet supplies a brief description of
the software you received with your PCI-DAS1002 and information regarding installation of that software.
Please read this booklet completely before installing any software or hardware.

Optional components
If you ordered any of the following products with your board, they should be included with your shipment.


C100FF-x cable11

PCI-DAS1002 User's ManualInstalling the PCI-DAS1002

Signal termination and conditioning accessories
MCC provides signal termination and conditioning products for use with the PCI-DAS1002. Refer to the"Field wiring and signal termination" section for a complete list of compatible accessory products.

Unpacking the PCI-DAS1002
As with any electronic device, you should take care while handling to avoid damage from static
electricity. Before removing the PCI-DAS1002 from its packaging, ground yourself using a wrist strap or bysimply touching the computer chassis or other grounded object to eliminate any stored static charge.
If any components are missing or damaged, notify Measurement Computing Corporation immediately byphone, fax, or e-mail:

Phone: 508-946-5100 and follow the instructions for reaching Tech Support.
Fax: 508-946-9500 to the attention of Tech Support
Email: techsupport@mccdaq.com

Installing the software
Refer to the Quick Start Guide for instructions on installing the software on the Measurement Computing Data
Acquisition Software CD. This booklet is available in PDF at www.mccdaq.com/PDFmanuals/DAQ-SoftwareQuick-Start.pdf.

Installing the hardware
The PCI-DAS1002 board is completely plug-and-play. There are no switches or jumpers to set on the board.
Configuration is controlled by your system's BIOS. To install your board, follow the steps below.
Install the MCC DAQ software before you install your board
The driver needed to run your board is installed with the MCC DAQ software. Therefore, you need to install the
MCC DAQ software before you install your board. Refer to the Quick Start Guide for instructions on installingthe software.
1.

Turn your computer off, open it up, and insert your board into an available PCI slot.

2.

Close your computer and turn it on.
If you are using an operating system with support for plug-and-play (such as Windows 2000 or
Windows XP), a dialog box opens as the system loads, indicating that new hardware has been detected. The
information file for this board should have already been loaded onto your PC when you installed the
Measurement Computing Data Acquisition Software CD supplied with your board, and should be detected
automatically by Windows. If you have not installed this software, cancel the dialog and install it now.

3.

To test your installation and configure your board, run the InstaCal utility installed in the previous section.
Refer to the Quick Start Guide that came with your board for information on how to initially set up andload InstaCal.

Allow your computer to warm up for at least 15 minutes before acquiring data with this board. The high speed
components used on the board generates heat, and it takes this amount of time for a board to reach steady state
if it has been powered off for a significant amount of time.

12

PCI-DAS1002 User's Manual

Installing the PCI-DAS1002

Configuring the hardware
All hardware configuration options on the PCI-DAS1002 are software controlled. You can select some of theconfiguration options using InstaCal, such as the analog input configuration (16 single-ended or eight
differential channels), and the source for the two independent counters. Once selected, any program that uses
the Universal Library will initialize the hardware according to these selections.
For general information regarding analog signal connection and configuration, refer to the Guide to Signal
Connections (available on our web site at www.mccdaq.com/signals/signals.pdf).

Differential input mode
When all channels are configured for differential input mode, eight analog input channels are available. In thismode, the input signal is measured with respect to the low input. The input signal is delivered through threewires:
The wire carrying the signal to be measured connects to CH# IN HI.
The wire carrying the reference signal connects to CH# IN LO.
The third wire, typically a system ground, connects to LLGND.

Single-ended input mode
When all channels are configured for single-ended input mode, 16 analog input channels are available. In thismode, the input signal is referenced to the board's signal ground (LLGND). The input signal is delivered
through two wires:

The wire carrying the signal to be measured connects to CH# IN HI.
The second wire connects to LLGND.

Connecting the board for I/O operations
Connectors, cables – main I/O connector
The table below lists the board connectors, applicable cables, and compatible accessory boards for the PCIDAS1002.
Board connectors, cables, accessory equipment
Connector typeCompatible cables
Compatible accessory products(with the C100FF-x cable)

100 pin high density Robinson Nugent connector
C100FF-x, unshielded ribbon cable. x = length in feet (Figure 2 on on page 16)
ISO-RACK16/P
ISO-DA02/P
CIO-ERB08*
CIO-ERB24*
SSR-RACK08*
SSR-RACK24*
BNC-16SE
BNC-16DI
CIO-MINI50 (2 required)
CIO-TERM100 (1 required)
SCB-50 (1 required)* These devices require the DADP-5037 PCI-DAS to 37-pin SSR and ERB adapter board.

13

PCI-DAS1002 User's Manual

Installing the PCI-DAS1002

Pinout – main I/O connector
8-channel differential mode
Signal Name
GND
N/C
N/C
N/C
N/C
A/D Int. Pacer Output
N/C
N/C
PC -12VPC GND
PC +12VPC GND
N/C
CTR5 OUT
CTR5 GATE
CTR5 CLK
N/C
N/C
CTR6 OUT
CTR6 GATECTR6 CLK
N/C
N/C
N/C
N/C
N/C
FIRSTPORTC Bit 7
FIRSTPORTC Bit 6
FIRSTPORTC Bit 5
FIRSTPORTC Bit 4
FIRSTPORTC Bit 3
FIRSTPORTC Bit 2
FIRSTPORTC Bit 1
FIRSTPORTC Bit 0
FIRSTPORTB Bit 7
FIRSTPORTB Bit 6
FIRSTPORTB Bit 5
FIRSTPORTB Bit 4
FIRSTPORTB Bit 3
FIRSTPORTB Bit 2
FIRSTPORTB Bit 1
FIRSTPORTB Bit 0
FIRSTPORTA Bit 7
FIRSTPORTA Bit 6
FIRSTPORTA Bit 5
FIRSTPORTA Bit 4
FIRSTPORTA Bit 3
FIRSTPORTA Bit 2
FIRSTPORTA Bit 1
FIRSTPORTA Bit 0
PCI slot

Pin
100
99
98
97
96
95
94
93
92
91
90
89
88
87
86
85
84
83
82
81
80
79
78
77
76
75
74
73
72
71
70
69
68
67
66
65
64
63
62
61
60
59
58
57
56
55
54
53
52
51

Pin
50
49
48
47
46
45
44
43
42
41
40
39
38
37
36
35
34
33
32
31
30
29
28
27
26
25
24
23
22
21
20
19
18
17
16
15
14
13
12
11
10
9
8
7
6
5
4
3
2
1

↓

14

Signal Name
GND
N/C
PC +5V
N/C
N/C
A/D External Trigger
IN
N/C
N/C
A/D External Pacer
CTR4 OUT
CTR4 GATECTR4 CLK
D/A OUT 1
D/A GND 1
D/A OUT 0
D/A GND 0
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
LLGND
CH7 IN LO
CH7 IN HI
CH6 IN LO
CH6 IN HI
CH5 IN LO
CH5 IN HI
CH4 IN LO
CH4 IN HI
CH3 IN LO
CH3 IN HI
CH2 IN LO
CH2 IN HI
CH1 IN LO
CH1 IN HI
CH0 IN LO
CH0 IN HI
LLGND

PCI-DAS1002 User's Manual

Installing the PCI-DAS1002

16-channel single-ended mode
Signal Name
GND
N/C
N/C
N/C
N/C
A/D Int. Pacer Output
N/C
N/C
PC -12VPC GND
PC +12VPC GND
N/C
CTR5 OUT
CTR5 GATE
CTR5 CLK
N/C
N/C
CTR6 OUT
CTR6 GATECTR6 CLK
N/C
N/C
N/C
N/C
N/C
FIRSTPORTC Bit 7
FIRSTPORTC Bit 6
FIRSTPORTC Bit 5
FIRSTPORTC Bit 4
FIRSTPORTC Bit 3
FIRSTPORTC Bit 2
FIRSTPORTC Bit 1
FIRSTPORTC Bit 0
FIRSTPORTB Bit 7
FIRSTPORTB Bit 6
FIRSTPORTB Bit 5
FIRSTPORTB Bit 4
FIRSTPORTB Bit 3
FIRSTPORTB Bit 2
FIRSTPORTB Bit 1
FIRSTPORTB Bit 0
FIRSTPORTA Bit 7
FIRSTPORTA Bit 6
FIRSTPORTA Bit 5
FIRSTPORTA Bit 4
FIRSTPORTA Bit 3
FIRSTPORTA Bit 2
FIRSTPORTA Bit 1
FIRSTPORTA Bit 0
PCI slot

Pin
100
99
98
97
96
95
94
93
92
91
90
89
88
87
86
85
84
83
82
81
80
79
78
77
76
75
74
73
72
71
70
69
68
67
66
65
64
63
62
61
60
59
58
57
56
55
54
53
52
51

Pin
50
49
48
47
46
45
44
43
42
41
40
39
38
37
36
35
34
33
32
31
30
29
28
27
26
25
24
23
22
21
20
19
18
17
16
15
14
13
12
11
10
9
8
7
6
5
4
3
2
1

↓

15

Signal Name
GND
N/C
PC +5V
N/C
N/C
A/D External Trigger In
N/CN/C
A/D External Pacer
CTR4 OUT
CTR4 GATECTR4 CLK
D/A OUT 1
D/A GND 1
D/A OUT 0
D/A GND 0
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
LLGND
CH15 IN
CH7 IN
CH14 IN
CH6 IN
CH13 IN
CH5 IN
CH12 IN
CH4 IN
CH11 IN
CH3 IN
CH10 IN
CH2 IN
CH9 IN
CH1 IN
CH8 IN
CH0 INLLGND

PCI-DAS1002 User's Manual

Installing the PCI-DAS1002

CablesCable is labeled“Pins 1-50”.

50

49

50

1

2

Key

100

The red stripe
identifies pin # 1
99 100

1

51

Key

51

Cable is labeled“Pins 51-100”.

52

The red stripe
identifies pin # 51

Figure 2. C100FF-x cable

Field wiring and signal termination
The table below lists the MCC screw terminal and signal conditioning boards that are compatible with the
PCI-DAS1000.
Screw terminal and
BNC adapters

http://www.mccdaq.com/products/accessories.aspx

SCB-50
CIO-MINI50
BNC-16SE
BNC-16DI

50-conductor, shielded signal connection box.
50-pin universal screw terminal accessory.
16-channel single-ended BNC connector box.
Eight-channel differential BNC connector box.
http://www.mccdaq.com/products/signal_conditioning.aspx

Signal conditioning

ISO-RACK16/P
16-channel isolation module mounting rack.
ISO-DA02/P
Two-channel 5B module rack.
SSR-RACK24*
24-position solid state relay rack.
SSR-RACK08*
Eight-channel solid state relay rack.
CIO-ERB24*
24-channel electromechanical relay accessory for digital I/O boards.
CIO-ERB08*
Eight-channel electromechanical relay accessory for digital I/O boards.
* These devices require the DADP-5037 PCI-DAS to 37-pin SSR and ERB adapter board.

16

Chapter 3

Calibrating the Board
Introduction
Calibrate the board using the InstaCal utility after the board has fully warmed up. The recommended warm-up
time is 15 minutes. For best results, calibrate the board immediately before making critical measurements. The
high resolution analog components on the board are sensitive to temperature. Pre-measurement calibration
ensures that your board is operating at optimum calibration values.

Calibration theory
The PCI-DAS1002 is shipped fully calibrated from the factory with calibration coefficients stored in nvRAM.
When using Universal Library at run time, these calibration factors are loaded into system memory and areautomatically retrieved each time a different DAC/ADC range is specified.
The analog input circuits are calibrated for both offset and gain (Figure 3).
Offset calibration for the analog input is performed directly on the input amplifier (PGIA) with coarse andfine trim DACs acting on the amplifier.
For input gain calibration, a precision calibration reference is used with coarse and fine trim DACs acting
on the ADC.
Analog In

Trim DAC
CoarseVariable
Gain

PGIA

A/D

Pre-Gain
OffsetTrim DAC
Fine
Figure 3. Analog input circuit calibration

The analog output circuits are calibrated for both offset and gain (Figure 4). This circuit is duplicated for both
DAC0 and DAC1.


Offset adjustments are made in the output buffer section. The tuning range of this adjustment allows for
max DAC and output buffer offsets.
Gain calibration is performed via DAC reference adjustments.
Analog Out

D/A
Ref

Trim DAC
CoarseGain
AdjustTrim DAC

OffsetAdjustTrim DAC
Fine

Figure 4. Analog output circuit calibration

17

Chapter 4

Specifications
Typical for 25 °C unless otherwise specified.
Specifications in italic text are guaranteed by design.

Analog input
Table 1. Analog input specifications
Parameter

Specification

A/D converter type
Resolution
Number of channels
Input ranges

ADS7800 or equivalent
12 bits
16 single-ended / 8 differential, software selectable±10 V, ±5 V, ±2.5 V, ±1.25 V, 0 to 10 V, 0 to 5V, 0 to 2.5 V, 0 to 1.25 V software
programmable
Internal counter - 82C54.
External source (A/D External Pacer)
software programmable for rising or falling edge
Software polled
External edge trigger (A/D EXTERNAL TRIGGER)
Rising edge trigger
Unlimited pre- and post-trigger samples. Total number of samples must be greater than
512.
Software selectable option, burst rate = 150 kHz
From 1024 sample FIFO via REPINSW
Programmed I/O
3 µs max
6 µs for a full scale step to 1 LSB
200 kHz
Auto-calibration, calibration factors for each range stored on board in nonvolatile RAM±0.25 LSB/°C max±0.25 LSB/°C max±10 V min-70 dB min±20 nA max
10 MOhms min±35 volts

A/D pacing(software programmable)

A/D trigger sources
A/D triggering modes
A/D pre-trigger mode
Burst mode
Data transfer
A/D conversion time
Analog front end settling time
Calibrated throughput
Calibration
A/D full-scale gain drift
A/D zero drift
Common mode range
CMRR @ 60 Hz
Input leakage current
Input impedance
Absolute maximum input
voltageWarm-up time

15 minutes

18

PCI-DAS1002 User's Manual

Specifications

Accuracy
Accuracies are listed for a 200 kHz sampling rate, 100 sample average, single channel operation, a 15 minutewarm-up, and operational temperatures within ±2 C of internal calibration temperature. The calibrator test
source high side is tied to Channel 0 In and the low side is tied to AGND.
Table 2. Absolute accuracy specifications (analog input)
Range

Absolute Accuracy±10.00 V±5.000 V±2.500 V±1.250 V
0 to 10.00 V
0 to 5.000 V
0 to 2.500 V
0 to 1.250 V±2.5 LSB max±2.5 LSB max±2.5 LSB max±2.5 LSB max±2.5 LSB max±2.5 LSB max±2.5 LSB max±2.5 LSB max

Each PCI-DAS1002 is tested at the factory to assure the board’s overall error does not exceed accuracy limits
described in Table 2 above.
Table 3. Calibrated accuracy components (in LSB)
Range

Gain Error

Offset Error

DLE

ILE

All ranges±1.0 max±1.0 max±0.75 max

±0.5 max

As shown in Table 3, total board error is a combination of gain, offset, differential linearity and integral
linearity error. The theoretical worst-case error of the board may be calculated by summing these component
errors. Worst case errors are realized only in the unlikely event that each of the component errors are at theirmaximum level, and causing error in the same direction.

Crosstalk
Crosstalk is defined here as the influence of one channel upon another when scanning two channels at the
maximum rate. A full scale 100 Hz triangle wave is input on channel 1; Channel 0 is tied to analog ground at
the 100-pin user connector. The table below summarizes the influence of channel 1 on channel 0 with theeffects of noise removed. The residue on channel zero is described in LSB’s.
Table 4. Crosstalk specifications
Condition

Crosstalk

Per channel Rate

ADC Rate

All ranges

2 LSB pk-pk

100 kHz200 kHz

Noise performance
Table 5 below summarizes the noise performance for the PCI-DAS1002. Noise distribution is determined bygathering 50 K samples at 200 kHz with inputs tied to ground at the user connector.
Table 5. Board noise performanceRange% within ±2 LSB% within ±1 LSB

LSBs

LSBrms*0 to 1.250 V
All other ranges

100%
100%

99%
100%

4
3

0.61
0.45

* RMS noise is defined as the peak-to-peak bin spread divided by 6.6.

19

PCI-DAS1002 User's Manual

Specifications

Analog output
Table 6. Analog output specifications
D/A converter type
Resolution
Number of channels
Configuration
Output range
D/A pacing
Data transfer
Monotonicity
Overall analog output drift
Settling timeSlew rate
Current drive
Output short-circuit duration
Output coupling
Output impedance
Miscellaneous

AD7847AR or equivalent
12 bits2
Voltage output, single-ended±10 V, ±5 V, 0 to 10 V, or 0 to 5 V. Software selectable. Each channel independently
programmable.
Software
Programmed I/O
Guaranteed monotonic over temperature±0.03 LSB/°C max
4 µs to 0.01% of 10 V step
7 V/µs min±5 mA minIndefinite @25 mA
DC
0.1 Ohm, max
Double buffered output latches
Output voltage on power up and reset: ±200 mV

Absolute accuracy
Table 7. Absolute accuracy specifications (analog output)
All ranges±3 LSB

Calibrated accuracy components
Table 8. Calibrated accuracy specifications (analog output)
Gain error
Offset error
Integral linearity error
Differential linearity error±1.0 LSB max±0.5LSB max±1.0 LSB max±1.0 LSB max

Each PCI-DAS1002 is tested at the factory to assure absolute accuracy.
Total analog output error is a combination of gain, offset, integral linearity and differential linearity error. The
theoretical worst-case error of the board may be calculated by summing these component errors. Worst caseerror is realized only in the unlikely event that each of the component errors are at their maximum level, andcausing error in the same direction. Although an examination of the chart and a summation of the maximum
theoretical errors shows that the board could theoretically exhibit a ±3.5 LSB error, our testing assures this error
is never realized in a board that we ship.

20

PCI-DAS1002 User's Manual

Specifications

Counters
Table 9. Counter specifications
Parameter

Specification

Counter type
Configuration
82C54A:Counter 0 - ADC residual
sample counter

82C54
Two 82C54 devices. 3 down counters per 82C54, 16 bits each
Source:
ADC Clock
Gate:
Internal programmable source
Output:End-of-Acquisition interrupt
Source:10 MHz oscillator
Gate:
Tied to counter 2 gate, programmable source
Output:Chained to counter 2 clock
Source:Counter 1 output
Gate:
Tied to counter 1 gate, programmable source
Output:
ADC pacer clock (if software selected). Available atuser connector.
Source:
ADC clock
Gate:
External trigger
Output:End-of-Acquisition interrupt
Source:User input at 100-pin connector (CTR4 CLK)or internal 10 MHz (software selectable)
Gate:
User input at 100-pin connector (CTR4 GATE)
Output:Available at 100-pin connector (CTR4 OUT)
Source:User input at 100-pin connector (CTR5 CLK)
Gate:
User input at 100-pin connector (CTR5 GATE)
Output:Available at 100-pin connector (CTR5 OUT)
Source:User input at 100-pin connector (CTR6 CLK)
Gate:
User input at 100-pin connector (CTR6 GATE)
Output:Available at 100-pin connector (CTR6 OUT)
10 MHz max
30 ns min
50 ns min
50 ns min
50 ns min
0.8 V max
2.0 V min
0.4 V max
3.0 V min
10 MHz
50 ppmCounter 1 - ADC pacer lower
dividerCounter 2 - ADC pacer upperdivider

82C54B:Counter 0 - pretrigger mode

Counter 0 - user counter 4(when in non-pretrigger mode)

Counter 1 - user counter 5Counter 2 - user counter 6

Clock input frequency
High pulse width (clock input)
Low pulse width (clock input)
Gate width highGate width low
Input low voltage
Input high voltage
Output low voltage
Output high voltage
Crystal oscillator frequency
Frequency accuracy

21

PCI-DAS1002 User's Manual

Specifications

Digital input/output
Table 10. DIO specifications
Digital type
Number of I/O
Configuration

Input high voltage
Input low voltage
Output high voltage (IOH = -2.5 mA)
Output low voltage (IOL = 2.5 mA)
Power-up / reset state

82C55
24 (FIRSTPORTA Bit 0 through FIRSTPORTC Bit 7)
2 banks of 8 and 2 banks of 4 or
3 banks of 8 or
2 banks of 8 with handshake
2.0 V min, 5.5 V absolute max
0.8 V max, –0.5 V absolute min
3.0 V min
0.4 V max
Input mode (high impedance)

Interrupts
Table 11. Interrupt specifications
Interrupts

PCI INTA# – mapped to IRQn via PCI BIOS at boot-time

Interrupt enable
Interrupt sources

Programmable. Default = disabled.
Residual sample counter
A/D End-of-channel-scan
A/D FIFO-not-empty
A/D FIFO-half-full
A/D Pacer

Miscellaneous
Table 12. Micellaneous specifications+5 Volts+12 Volts-12 Volts

Available at I\O connector (PC +5V)
Available at I\O connector (PC +12V)
Available at I\O connector (PC –12V)

Power consumption
Table 13. Power consumption specifications+5 V

0.8 A typical, 1.0 A max

Environmental
Table 14. Environmental specifications
Operating temperature range
Storage temperature range
Humidity

0 to 70 °C-40 to 100 °C
0 to 95% non-condensing

Mechanical
Table 15. Mechanical specifications
Card dimensions

PCI half card: 174.63 mm (L) x 106.86 mm (H) x 14.48 mm (D)22

PCI-DAS1002 User's Manual

Specifications

Connector and pin out
Table 16. Main connector specifications
Connector typeCompatible cableCompatible accessory products(with C100FF-x cable)

100-pin high-density Robinson Nugent
C100FF-x, unshielded ribbon cable. x = length in feet
ISO-RACK16/P
ISO-DA02/P
CIO-ERB24 (DADP-5037 adaptor required)
CIO-SERB24/FD (DADP-5037 adaptor required)
SSR-RACK24 (DADP-5037 adaptor required)BNC-16SE
BNC-16DI
CIO-MINI50 (2 required)
CIO-TERM100 (1 required)
SCB-50 (1 required)8-channel differential mode pin out
Pin
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18
19
20
21
22
23
24
25
26
27
28
29
30
31
32
33
34
35
36
37
38
39
40
41
42
43
44
45
46
47
48
49
50

Signal Name
LLGND
CH0 HI
CH0 LOCH1 HI
CH1 LOCH2 HI
CH2 LOCH3 HI
CH3 LOCH4 HI
CH4 LOCH5 HI
CH5 LOCH6 HI
CH6 LOCH7 HI
CH7 LO
LLGND
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
D/A GND 0
D/A OUT 0
D/A GND 1
D/A OUT 1
CTR4 CLK
CTR4 GATECTR4 OUT
A/D EXTERNAL PACER
N/C
N/C
A/D EXTERNAL TRIGGER IN
N/C
N/C
PC +5V
N/C
GND

Pin
51
52
53
54
55
56
57
58
59
60
61
62
63
64
65
66
67
68
69
70
71
72
73
74
75
76
77
78
79
80
81
82
83
84
85
86
87
88
89
90
91
92
93
94
95
96
97
98
99
100

23

Signal Name
FIRSTPORTA Bit 0
FIRSTPORTA Bit 1
FIRSTPORTA Bit 2
FIRSTPORTA Bit 3
FIRSTPORTA Bit 4
FIRSTPORTA Bit 5
FIRSTPORTA Bit 6
FIRSTPORTA Bit 7
FIRSTPORTB Bit 0
FIRSTPORTB Bit 1
FIRSTPORTB Bit 2
FIRSTPORTB Bit 3
FIRSTPORTB Bit 4
FIRSTPORTB Bit 5
FIRSTPORTB Bit 6
FIRSTPORTB Bit 7
FIRSTPORTC Bit 0
FIRSTPORTC Bit 1
FIRSTPORTC Bit 2
FIRSTPORTC Bit 3
FIRSTPORTC Bit 4
FIRSTPORTC Bit 5
FIRSTPORTC Bit 6
FIRSTPORTC Bit 7
N/C
N/C
N/C
N/C
N/C
CTR6 CLKCTR6 GATECTR6 OUT
N/C
N/C
CTR5 CLK
CTR5 GATE
CTR5 OUT
N/C
GND
+12V
GND
-12V
N/C
N/C
A/D INTERNAL PACER OUTPUT
N/C
N/C
N/C
N/C
GND

PCI-DAS1002 User's Manual

Specifications

16-channel single-ended mode pin out
Pin
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18
19
20
21
22
23
24
25
26
27
28
29
30
31
32
33
34
35
36
37
38
39
40
41
42
43
44
45
46
47
48
49
50

Signal Name
LLGND
CH0 HI
CH8 HICH1 HI
CH9 HICH2 HI
CH10 HI
CH3 HI
CH11 HI
CH4 HI
CH12 HICH5 HI
CH13 HICH6 HI
CH14 HI
CH7 HI
CH15 HI
LLGND
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
N/C
D/A GND 0
D/A OUT 0
D/A GND 1
D/A OUT 1
CTR4 CLK
CTR4 GATECTR4 OUT
A/D EXTERNAL PACER
N/C
N/C
A/D EXTERNAL TRIGGER IN
N/C
N/C
PC +5V
N/C
GND

Pin
51
52
53
54
55
56
57
58
59
60
61
62
63
64
65
66
67
68
69
70
71
72
73
74
75
76
77
78
79
80
81
82
83
84
85
86
87
88
89
90
91
92
93
94
95
96
97
98
99
100

24

Signal Name
FIRSTPORTA Bit 0
FIRSTPORTA Bit 1
FIRSTPORTA Bit 2
FIRSTPORTA Bit 3
FIRSTPORTA Bit 4
FIRSTPORTA Bit 5
FIRSTPORTA Bit 6
FIRSTPORTA Bit 7
FIRSTPORTB Bit 0
FIRSTPORTB Bit 1
FIRSTPORTB Bit 2
FIRSTPORTB Bit 3
FIRSTPORTB Bit 4
FIRSTPORTB Bit 5
FIRSTPORTB Bit 6
FIRSTPORTB Bit 7
FIRSTPORTC Bit 0
FIRSTPORTC Bit 1
FIRSTPORTC Bit 2
FIRSTPORTC Bit 3
FIRSTPORTC Bit 4
FIRSTPORTC Bit 5
FIRSTPORTC Bit 6
FIRSTPORTC Bit 7
N/C
N/C
N/C
N/C
N/C
CTR6 CLKCTR6 GATECTR6 OUT
N/C
N/C
CTR5 CLK
CTR5 GATE
CTR5 OUT
N/C
GND
+12V
GND
-12V
N/C
N/C
A/D INTERNAL PACER OUTPUT
N/C
N/C
N/C
N/C
GND

Declaration of Conformity
Manufacturer:
Address:

Category:

Measurement Computing Corporation
10 Commerce Way
Suite 1008
Norton, MA 02766
USA

Information technology equipment.

Measurement Computing Corporation declares under sole responsibility that the product

PCI-DAS1002
to which this declaration relates is in conformity with the relevant provisions of the following standards or otherdocuments:
EC EMC Directive 2004/108/EC: Electromagnetic Compatibility, EN 61326-1:2006 (IEC 61326-1:2005)
Emissions: Group 1, Class B


EN55022 (1995)/CISPR 22: Radiated and Conducted emissions.

Immunity: EN61326-1:2006, (IEC 61326-1:2005)







EN61000-4-2 (2001): Electrostatic Discharge immunity.
EN61000-4-3 (2002): Radiated Electromagnetic Field immunity.
EN61000-4-4 (2004): Electric Fast Transient Burst immunity.
EN61000-4-5 (2001): Surge immunity.
EN61000-4-6 (2003): Radio Frequency Common Mode immunity.
EN61000-4-11 (2004): Voltage Dip and Interrupt immunity.

Declaration of Conformity based on tests conducted by Chomerics Test Services, Woburn, MA 01801, USA in
September, 2001. Test records are outlined in Chomerics Test Report #EMI3053.01. Further testing was
conducted by Chomerics Test Services, Woburn, MA. 01801, USA in December, 2008. Test records are
outlined in Chomerics Test report #EMI5241.08.
We hereby declare that the equipment specified conforms to the above Directives and Standards.

Carl Haapaoja, Director of Quality Assurance

Measurement Computing Corporation
10 Commerce Way
Suite 1008
Norton, Massachusetts 02766(508) 946-5100
Fax: (508) 946-9500
E-mail: info@mccdaq.com
www.mccdaq.com