6K-SSR-RACK08
8-solid state relay accessory board for use
with PCI-DAS6000 Series and
PCI-DAC6700 Series boards
User's Guide

Document Revision 2A, July, 2006
© Copyright 2006, Measurement Computing Corporation

Trademark and Copyright Information
Measurement Computing Corporation, InstaCal, Universal Library, and the Measurement Computing logo areeither trademarks or registered trademarks of Measurement Computing Corporation. Refer to the Copyrights &Trademarks section on mccdaq.com/legal for more information about Measurement Computing trademarks.
Other product and company names mentioned herein are trademarks or trade names of their respective
companies.
© 2006 Measurement Computing Corporation. All rights reserved. No part of this publication may be
reproduced, stored in a retrieval system, or transmitted, in any form by any means, electronic, mechanical, by
photocopying, recording, or otherwise without the prior written permission of Measurement Computing
Corporation.
NoticeMeasurement Computing Corporation does not authorize any Measurement Computing Corporation product for
use in life support systems and/or devices without prior written consent from Measurement Computing
Corporation. Life support devices/systems are devices or systems that, a) are intended for surgical implantation
into the body, or b) support or sustain life and whose failure to perform can be reasonably expected to result ininjury. Measurement Computing Corporation products are not designed with the components required, and are
not subject to the testing required to ensure a level of reliability suitable for the treatment and diagnosis ofpeople.

HM 6K-SSR-RACK08.doc

ii

iii

Table of Contents
Preface

About this User's Guide ......................................................................................................................vi
What you will learn from this user's guide ........................................................................................................vi
Conventions in this user's guide ........................................................................................................................vi
Where to find more information ........................................................................................................................vi
Chapter 1

Introducing the 6K-SSR-RACK08 .................................................................................................... 1-1
Overview: 6K-SSR-RACK08 features ........................................................................................................... 1-1
Software features ............................................................................................................................................ 1-1
Chapter 2

Installing the 6K-SSR-RACK08 ........................................................................................................ 2-1
What comes with your 6K-SSR-RACK08 shipment? .................................................................................... 2-1
Hardware ....................................................................................................................................................................... 2-1
Additional documentation.............................................................................................................................................. 2-2
Optional components ..................................................................................................................................................... 2-2

Unpacking the 6K-ERB08.............................................................................................................................. 2-2
Connecting the 6K-SSR-RACK08 for I/O operations.................................................................................... 2-2
Connectors, cables – interface I/O connectors ............................................................................................................... 2-3
Pin out – interface connectors........................................................................................................................................ 2-4Signal cables .................................................................................................................................................................. 2-7

Connecting to your 6000 Series control board ............................................................................................... 2-8
Configuring the 6K-SSR-RACK08 board ...................................................................................................... 2-8
Power source.................................................................................................................................................................. 2-9
I/O type ........................................................................................................................................................................ 2-10
Relay logic ................................................................................................................................................................... 2-10
Chapter 3

Functional Details ............................................................................................................................. 3-1
Components.................................................................................................................................................... 3-2
Compatible SSR modules .............................................................................................................................................. 3-3

Screw terminal connections............................................................................................................................ 3-3
Wire gauge..................................................................................................................................................................... 3-3Signal level control and power up conditions................................................................................................. 3-3
LED states ...................................................................................................................................................... 3-4
6K-SSR-RACK08 assembly diagram............................................................................................................. 3-5
Chapter 4

Specifications.................................................................................................................................... 4-1
Power consumption ........................................................................................................................................ 4-1
Environmental ................................................................................................................................................ 4-1
Mechanical ..................................................................................................................................................... 4-1
Relay screw terminals..................................................................................................................................... 4-1
Power in jumper (JP1) .................................................................................................................................... 4-2
I/O module type selection (JP2, JP3).............................................................................................................. 4-2
I/O module polarity selection (JP5, JP6) ........................................................................................................ 4-2
Bypass resistors .............................................................................................................................................. 4-2
Compatible products....................................................................................................................................... 4-2

iv

6K-SSR-RACK08 User's Guide

Main connectors and pin out........................................................................................................................... 4-3
User connector P11........................................................................................................................................................ 4-3
User connector P12........................................................................................................................................................ 4-4
User connector P9.......................................................................................................................................................... 4-5
User connector P10........................................................................................................................................................ 4-6v

Preface

About this User's Guide
What you will learn from this user's guideThis user’s guide explains how to install, configure, and use the 6K-SSR-RACK08 so that you get the most outof its features.
This user’s guide also refers you to related documents available on our web site, and to technical support
resources.

Conventions in this user's guide
For more information on …
Text presented in a box signifies additional information and helpful hints related to the subject matter you arereading.
Caution! Shaded caution statements present information to help you avoid injuring yourself and others,
damaging your hardware, or losing your data.
<#:#>

Angle brackets that enclose numbers separated by a colon signify a range of numbers, such as those assigned
to registers, bit settings, etc.

bold text

Bold text is used for the names of objects on the screen, such as buttons, text boxes, and check boxes. For
example:1. Insert the disk or CD and click the OK button.

italic text

Italic text is used for the names of manuals and help topic titles, and to emphasize a word or phrase. For
example:The InstaCal installation procedure is explained in the Quick Start Guide.
Never touch the exposed pins or circuit connections on the board.

Where to find more information
The following electronic documents provide information that is relevant to the operation of the 6K-SSRRACK08.







MCC's Specifications: 6K-SSR-RACK08 (the PDF version of the Specifications chapter in this guide) isavailable on our web site at www.mccdaq.com/pdfs/SSR-RACK08.pdf.
MCC's Quick Start Guide is available on our web site atwww.mccdaq.com/PDFmanuals/DAQ-Software-Quick-Start.pdf.
MCC's Guide to Signal Connections is available on our web site atwww.mccdaq.com/signals/signals.pdf.
MCC's Universal Library User's Guide is available on our web site atwww.mccdaq.com/PDFmanuals/sm-ul-user-guide.pdf.
MCC's Universal Library Function Reference is available on our web site atwww.mccdaq.com/PDFmanuals/sm-ul-functions.pdf.
MCC's Universal Library for LabVIEW™ User’s Guide is available on our web site atwww.mccdaq.com/PDFmanuals/SM-UL-LabVIEW.pdf.

6K-SSR-RACK08 User's Guide (this document) is also available on our web site atwww.mccdaq.com/PDFmanuals/SSR-RACK08.pdf.

vi6K-SSR-RACK08 User's Guide

About this User's Guide

Links to the hardware manuals for each board supported by the 6K-SSR-RACK08 are provided below:
PCI-DAS6000 Series boards

Online User’s Guide

PCI-DAS6013, PCI-DAS6014
PCI-DAS6023, PCI-DAS6025
PCI-DAS6030, PCI-DAS6032
PCI-DAS6031, PCI-DAS6033
PCI-DAS6034, PCI-DAS6035, PCI-DAS6036
PCI-DAS6040
PCI-DAS6052
PCI-DAS6070
PCI-DAS6071

www.mccdaq.com/PDFmanuals/pci-das6013-6014.pdf
www.mccdaq.com/PDFmanuals/pci-das6025-23.pdf
www.mccdaq.com/PDFmanuals/pci-das6030-32.pdf
www.mccdaq.com/PDFmanuals/pci-das6031-33.pdf
www.mccdaq.com/PDFmanuals/pci-das603x.pdf
www.mccdaq.com/PDFmanuals/pci-das6040.pdf
www.mccdaq.com/PDFmanuals/pci-das6052.pdf
www.mccdaq.com/PDFmanuals/pci-das6070.pdf
www.mccdaq.com/PDFmanuals/pci-das6071.pdf

PCI-DAC6700 Series boards
PCI-DAC6702, PCI-DAC6703

www.mccdaq.com/PDFmanuals/pci-dac670x.pdf

viiChapter 1

Introducing the 6K-SSR-RACK08
Overview: 6K-SSR-RACK08 features
This manual explains how to install, configure, and use the 6K-SSR-RACK08 with supported Measurement
Computing hardware. The 6K-SSR-RACK08 is designed for use in your control applications to switch on andoff a variety of devices, such as fans, blowers, pumps, etc.
The 6K-SSR-RACK08 is a mounting rack for eight solid state relay (SSR) modules. The AUXPORT digital I/O
lines (DIO0 – DIO7) on your on your PCI-DAS6000 Series board or PCI-DAC6700 Series board directly
control the SSR modules on your 6K-SSR-RACK08. The PCI-DAS6000 Series board and PCI-DAC6700
Series board are referred to as the "6000 Series control board" for the remainder of this guide.
The 6K-SSR-RACK08 backplane is divided into two groups of four modules each. You can configure each
module group for input or output using an on-board jumper. You cannot mix input and output modules within agroup. The following configurations are possible.
four input and four output modules
eight input modules
eight output modules

You can set the relays for each module group for active high or low logic with an on-board jumper. Independent
LEDs at each module position indicate the on/off status of each module. The board has an open component
location where you can install a pull-up or pull-down resistor to pull the digital relay control line low when
disconnected from the 6000 Series control board or when the digital lines on the 6000 Series control board arein high impedance (or input) mode.
The 6K-SSR-RACK08 has eight independent screw terminal pairs for your field wiring connections. Each SSR
module has a dedicated positive and negative screw terminal.
You can power the 6K-SSR-RACK08 from your 6000 Series control board, from your computer's power
connectors, or from the AC adapter included with your 6K-SSR-RACK08.
The 6K-SSR-RACK08 has multiple interface connectors that you can use to daisy chain to other accessory
board types.

Software features
For information on the features of InstaCal and the other software included with your 6K-ERB08, refer to the
Quick Start Guide that shipped with your device. The Quick Start Guide is also available in PDF at
www.mccdaq.com/PDFmanuals/DAQ-Software-Quick-Start.pdf.
Check www.mccdaq.com/download.htm for the latest software version or versions of the software supported
under less commonly used operating systems.

1-1

Chapter 2

Installing the 6K-SSR-RACK08
What comes with your 6K-SSR-RACK08 shipment?
The following items are shipped with the 6K-SSR-RACK08.

Hardware

6K-SSR-RACK08 (shown with standoffs attached)C-PCPOWER-10 cable — 10 foot cable to connect with your computer’s +5 V power supply connectors or
to an external +5 V power supply.



CB-PWR-9 AC adapter — 9 volt, 1 amp DC power supply, 110 VAC input

Six (6) standoffs with screws for attaching to the 6K-SSR-RACK08 board
2-1

6K-SSR-RACK08 User's Guide

Installing the 6K-SSR-RACK08

Additional documentation
In addition to this hardware user's guide, you should also receive the Quick Start Guide (available in PDF at
www.mccdaq.com/PDFmanuals/DAQ-Software-Quick-Start.pdf). This booklet supplies a brief description of
the software you received with your 6K-ERB08 and information regarding installation of that software. Please
read this booklet completely before installing any software or hardware.

Optional components


Cables

C100MMS-x

C100HD50-x

C50FF-x

Unpacking the 6K-ERB08
As with any electronic device, you should take care while handling to avoid damage from static
electricity. Before removing the 6K-ERB08 from its packaging, ground yourself using a wrist strap or by simply
touching the computer chassis or other grounded object to eliminate any stored static charge.
If any components are missing or damaged, notify Measurement Computing Corporation immediately byphone, fax, or e-mail:

Phone: 508-946-5100 and follow the instructions for reaching Tech Support.
Fax: 508-946-9500 to the attention of Tech Support
Email: techsupport@mccdaq.com

Connecting the 6K-SSR-RACK08 for I/O operations
Follow the guidelines below when connecting the 6K-SSR-RACK08 to your computer and to your MCC
hardware.


Make sure your connections to the 6K-SSR-RACK08 do not exceed any of the specified maximum ratings,
or you could damage the 6K-SSR-RACK08, your computer, and your 6000 Series control board. The
maximum ratings for connections to the 6K-SSR-RACK08 are listed in Chapter 4 "Specifications."
 Turn off your PC and any 6K-SSR-RACK08 power sources before you connect or disconnect the cablefrom the 6K-SSR-RACK08 to the 6000 Series control board.
 Connect one of the 6K-SSR-RACK08 board’s interface connectors to the I/O connector on the 6000 Series
control board. Refer to Table 2-1 for appropriate cable and connector combinations. Use the remaining
interface connectors to connect with other board types, such as the CIO-MINI50 or CIO-TERM100 screw
terminal boards.
 Use the module type color coding chart on the 6K-SSR-RACK08 board as a guide when you install the
SSR modules.
External power is required when using the C100HD50-x cable
The C100HD50-x cable does not provide power to the 6K-SSR-RACK08. If you use the C100HD50-x cable,
power the 6K-SSR-RACK08 with an external power source.

2-2

6K-SSR-RACK08 User's Guide

Caution!

Installing the 6K-SSR-RACK08

Connect the 6K-SSR-RACK08 to a PCI-DAS6000 Series board or to a PCI-DAC6700 Series board
only. Connecting to a board that is not supported by the 6K-SSR-RACK08 can damage both boards.

Connectors, cables – interface I/O connectors
The 6K-SSR-RACK08 board’s eight SSR modules are controlled by your 6000 Series control board’s digitalchannels DIO<0:7> through AUXPORT. Connect your signal cable from the 6000 Series control board’s
interface connector to one of the interface connectors on the 6K-SSR-RACK08 board.
The 6K-SSR-RACK08 has two 100-pin interface connectors labeled P11 and P12, and two 50-pin interface
connectors labeled P9 and P10. Signals pass through each connector. Only connect one of the four interface
connectors to your 6000 Series control board.
Daisy chaining to other 6K-SSR-RACK08 boards
Do not daisy chain additional 6K-SSR-RACK08 boards to the extra interface connectors, since any additionalboards would be controlled with the same DIO lines.
Table 2-1 lists the board connectors, compatible cables, and compatible MCC hardware for the 6K-SSRRACK08.
Table 2-1. Board connectors, compatible cables and hardware
Connector typeCompatible cable(connect to P11 or P12)
Compatible cable(connect to P9 or P10)

Compatible MCC hardware P11 and P12: Shielded, SCSI 100-pin D-type P9 and P10: Unshielded 50 pin ribbon connector
C100MMS-x shielded round cable. (see Figure 2-1). x = 1, 2 or 3 meters C100HD50-x unshielded ribbon cable (see Figure 2-2). x = 3 or 6 feet
 C50FF-x unshielded ribbon cable (see Figure 2-3)
x = 1, 2, 3, 4, 5, 10, 15, 20, 25 or 50 feet
Note: These cables do not provide power to the 6K-SSR-RACK08. External power isrequired.
PCI-DAS6000 Series analog input boards:
 PCI-DAS6013
 PCI-DAS6036
 PCI-DAS6014
 PCI-DAS6052
 PCI-DAS6030
 PCI-DAS6023
 PCI-DAS6031
 PCI-DAS6025
 PCI-DAS6032
 PCI-DAS6040
 PCI-DAS6033
 PCI-DAS6070
 PCI-DAS6034
 PCI-DAS6071
 PCI-DAS6035
PCI-DAC6700 Series analog output boards:
 PCI-DAC6702
 PCI-DAC6703

2-3

6K-SSR-RACK08 User's Guide

Installing the 6K-SSR-RACK08

Pin out – interface connectors
Table 2-2. Connector P11 pin out
Signal Name

GND
P12 Pass Through 99
P12 Pass Through 98
P12 Pass Through 97
P12 Pass Through 96
P12 Pass Through 95
P12 Pass Through 94
P12 Pass Through 93
DIO7
DIO6
DIO5
DIO4
DIO3
DIO2
DIO1
DIO0
P12 Pass Through 84
P12 Pass Through 83
P12 Pass Through 82
P12 Pass Through 81
P12 Pass Through 80
P12 Pass Through 79
P12 Pass Through 78
P12 Pass Through 77
P12 Pass Through 76
P12 Pass Through 75
P12 Pass Through 74
P12 Pass Through 73
P12 Pass Through 72
P12 Pass Through 71
P12 Pass Through 70
P12 Pass Through 69
P12 Pass Through 68
P12 Pass Through 67
P12 Pass Through 66
P12 Pass Through 65
P12 Pass Through 64
P12 Pass Through 63
P12 Pass Through 62
P12 Pass Through 61
P12 Pass Through 60
P12 Pass Through 59
P12 Pass Through 58
P12 Pass Through 57
P12 Pass Through 56
P12 Pass Through 55
P12 Pass Through 54
P12 Pass Through 53
P12 Pass Through 52
P12 Pass Through 51

Pin

100
99
98
97
96
95
94
93
92
91
90
89
88
87
86
85
84
83
82
81
80
79
78
77
76
75
74
73
72
71
70
69
68
67
66
65
64
63
62
61
60
59
58
57
56
55
54
53
52
51

••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••

2-4

Pin
50
49
48
47
46
45
44
43
42
41
40
39
38
37
36
35
34
33
32
31
30
29
28
27
26
25
24
23
22
21
20
19
18
17
16
15
14
13
12
11
10
9
8
7
6
5
4
3
2
1

Signal Name

GND
P12 Pass Through 49
P12 Pass Through 48
P12 Pass Through 47
P12 Pass Through 46
P12 Pass Through 45
P12 Pass Through 44
P12 Pass Through 43
P12 Pass Through 42
P12 Pass Through 41
P12 Pass Through 40
PC +5VP12 Pass Through 38
P12 Pass Through 37
P12 Pass Through 36
P12 Pass Through 35
P12 Pass Through 34
P12 Pass Through 33
P12 Pass Through 32
P12 Pass Through 31
P12 Pass Through 30
P12 Pass Through 29
P12 Pass Through 28
P12 Pass Through 27
P12 Pass Through 26
P12 Pass Through 25
P12 Pass Through 24
P12 Pass Through 23
P12 Pass Through 22
P12 Pass Through 21
P12 Pass Through 20
P12 Pass Through 19
P12 Pass Through 18
P12 Pass Through 17
P12 Pass Through 16
P12 Pass Through 15
P12 Pass Through 14
P12 Pass Through 13
P12 Pass Through 12
P12 Pass Through 11
P12 Pass Through 10
P12 Pass Through 9
P12 Pass Through 8
P12 Pass Through 7
P12 Pass Through 6
P12 Pass Through 5
P12 Pass Through 4
P12 Pass Through 3
P12 Pass Through 2
P12 Pass Through 1

6K-SSR-RACK08 User's Guide

Installing the 6K-SSR-RACK08

Table 2-3. Connector P12 pin out
Signal Name

GND
P11 Pass Through 99
P11 Pass Through 98
P11 Pass Through 97
P11 Pass Through 96
P11 Pass Through 95
P11 Pass Through 94
P11 Pass Through 93
DIO7
DIO6
DIO5
DIO4
DIO3
DIO2
DIO1
DIO0
P11 Pass Through 84
P11 Pass Through 83
P11 Pass Through 82
P11 Pass Through 81
P11 Pass Through 80
P11 Pass Through 79
P11 Pass Through 78
P11 Pass Through 77
P11 Pass Through 76
P11 Pass Through 75
P11 Pass Through 74
P11 Pass Through 73
P11 Pass Through 72
P11 Pass Through 71
P11 Pass Through 70
P11 Pass Through 69
P11 Pass Through 68
P11 Pass Through 67
P11 Pass Through 66
P11 Pass Through 65
P11 Pass Through 64
P11 Pass Through 63
P11 Pass Through 62
P11 Pass Through 61
P11 Pass Through 60
P11 Pass Through 59
P11 Pass Through 58
P11 Pass Through 57
P11 Pass Through 56
P11 Pass Through 55
P11 Pass Through 54
P11 Pass Through 53
P11 Pass Through 52
P11 Pass Through 51

Pin

100
99
98
97
96
95
94
93
92
91
90
89
88
87
86
85
84
83
82
81
80
79
78
77
76
75
74
73
72
71
70
69
68
67
66
65
64
63
62
61
60
59
58
57
56
55
54
53
52
51

••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••

2-5

Pin
50
49
48
47
46
45
44
43
42
41
40
39
38
37
36
35
34
33
32
31
30
29
28
27
26
25
24
23
22
21
20
19
18
17
16
15
14
13
12
11
10
9
8
7
6
5
4
3
2
1

Signal Name

GND
P11 Pass Through 49
P11 Pass Through 48
P11 Pass Through 47
P11 Pass Through 46
P11 Pass Through 45
P11 Pass Through 44
P11 Pass Through 43
P11 Pass Through 42
P11 Pass Through 41
P11 Pass Through 40
PC +5VP11 Pass Through 38
P11 Pass Through 37
P11 Pass Through 36
P11 Pass Through 35
P11 Pass Through 34
P11 Pass Through 33
P11 Pass Through 32
P11 Pass Through 31
P11 Pass Through 30
P11 Pass Through 29
P11 Pass Through 28
P11 Pass Through 27
P11 Pass Through 26
P11 Pass Through 25
P11 Pass Through 24
P11 Pass Through 23
P11 Pass Through 22
P11 Pass Through 21
P11 Pass Through 20
P11 Pass Through 19
P11 Pass Through 18
P11 Pass Through 17
P11 Pass Through 16
P11 Pass Through 15
P11 Pass Through 14
P11 Pass Through 13
P11 Pass Through 12
P11 Pass Through 11
P11 Pass Through 10
P11 Pass Through 9
P11 Pass Through 8
P11 Pass Through 7
P11 Pass Through 6
P11 Pass Through 5
P11 Pass Through 4
P11 Pass Through 3
P11 Pass Through 2
P11 Pass Through 1

6K-SSR-RACK08 User's Guide

Installing the 6K-SSR-RACK08

Table 2-4. Connector P9 pin out
Signal Name

Pin

GND
P10 Pass Through 98
P10 Pass Through 96
P10 Pass Through 94
DIO7
DIO5
DIO3
DIO1
P10 Pass Through 84
P10 Pass Through 82
P10 Pass Through 80
P10 Pass Through 78
P10 Pass Through 76
P10 Pass Through 74
P10 Pass Through 72
P10 Pass Through 70
P10 Pass Through 68
P10 Pass Through 66
P10 Pass Through 64
P10 Pass Through 62
P10 Pass Through 60
P10 Pass Through 58
P10 Pass Through 56
P10 Pass Through 54
P10 Pass Through 52

50
48
46
44
42
40
38
36
34
32
30
28
26
24
22
20
18
16
14
12
10
8
6
4
2

••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••

Pin

Signal Name

49
47
45
43
41
39
37
35
33
31
29
27
25
23
21
19
17
15
13
11
9
7
5
3
1

P10 Pass Through 99
P10 Pass Through 97
P10 Pass Through 95
P10 Pass Through 93
DIO6
DIO4
DIO2
DIO0
P10 Pass Through 83
P10 Pass Through 81
P10 Pass Through 79
P10 Pass Through 77
P10 Pass Through 75
P10 Pass Through 73
P10 Pass Through 71
P10 Pass Through 69
P10 Pass Through 67
P10 Pass Through 65
P10 Pass Through 63
P10 Pass Through 61
P10 Pass Through 59
P10 Pass Through 57
P10 Pass Through 55
P10 Pass Through 53
P10 Pass Through 51

Table 2-5. Connector P10 pin out
Signal Name

Pin

GND
P9 Pass Through 98
P9 Pass Through 96
P9 Pass Through 94
DIO7
DIO5
DIO3
DIO1
P9 Pass Through 84
P9 Pass Through 82
P9 Pass Through 80
P9 Pass Through 78
P9 Pass Through 76
P9 Pass Through 74
P9 Pass Through 72
P9 Pass Through 70
P9 Pass Through 68
P9 Pass Through 66
P9 Pass Through 64
P9 Pass Through 62
P9 Pass Through 60
P9 Pass Through 58
P9 Pass Through 56
P9 Pass Through 54
P9 Pass Through 52

50
48
46
44
42
40
38
36
34
32
30
28
26
24
22
20
18
16
14
12
10
8
6
4
2

••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••

Pin

Signal Name

49
47
45
43
41
39
37
35
33
31
29
27
25
23
21
19
17
15
13
11
9
7
5
3
1

P9 Pass Through 99
P9 Pass Through 97
P9 Pass Through 95
P9 Pass Through 93
DIO6
DIO4
DIO2
DIO0
P9 Pass Through 83
P9 Pass Through 81
P9 Pass Through 79
P9 Pass Through 77
P9 Pass Through 75
P9 Pass Through 73
P9 Pass Through 71
P9 Pass Through 69
P9 Pass Through 67
P9 Pass Through 65
P9 Pass Through 63
P9 Pass Through 61
P9 Pass Through 59
P9 Pass Through 57
P9 Pass Through 55
P9 Pass Through 53
P9 Pass Through 51

Information on signal connections
General information regarding signal connection and configuration is available in the Guide to Signal
Connections on our web site at www.mccdaq.com/signals/signals.pdf.

2-6

6K-SSR-RACK08 User's Guide

Installing the 6K-SSR-RACK08

Signal cables
50

100

50

100

1

51

1

51

Figure 2-1. C100MMS-x cable

For more information on the C100MMS-x cable, go to www.mccdaq.com
.
Pins 1-50 are on the long sideof the “D” connector.

50

Strain relief isstamped “Pins 1-50”.

49 50

100

Key
2

1

The red stripe
identifies pin # 1
99 100

1

51

Key

Pins 51-100 are on
the short side of
the “D” connector.

51 52

Strain relief is
Stamped “Pins 51-100”.

The red stripe
identifies pin # 51

Figure 2-2. C100HD50-x cable

For more information on the C100HD50-x cable, go to www.mccdaq.com
.
2

1

50

49

The red stripe
identifies pin # 1

Figure 2-3. C50FF-x cable

For more information on the C50FF-x cable, go to www.mccdaq.com
.

2-7

2

1

50

49

6K-SSR-RACK08 User's Guide

Installing the 6K-SSR-RACK08

Connecting to your 6000 Series control board
Perform the following procedure to connect the 6K-SSR-RACK08 to the 6000 Series control board.
1.

Turn off your computer and open the cover.

2.

Install your PCI-DAS6000 Series or PCI-DAC6700 Series board in the computer.

3.

Connect the digital signal cable from one of the interface connectors on the 6K-SSR-RACK08 to the
100-pin I/O connector on the 6000 Series control board.
oC100MMS-x cable: Connect one end to either 100-pin interface connector (P12 or P11) on the 6KSSR-RACK08. Connect the other end to the 100-pin interface connector on the 6000 Series control
board.

o

C100HD50-x cable: Connect the 50-pin end (stamped Pins 51 to 100) to either 50-pin connector (P9or P10) on the 6K-SSR-RACK08. Connect the 100-pin cable end to the 100-pin interface connector on
the 6000 Series control board.

External power is required when using the C100HD50-x cable
The C100HD50-x cable does not provide power to the 6K-SSR-RACK08. If you use the C100HD50-x cable,
power the 6K-SSR-RACK08 with an external power source.
4.

To connect other accessory boards or devices to your 6K-SSR-RACK08, use the remaining 100-pin
interface connector and/or the two 50-pin connectors for the connection(s). The following typicalconfigurations are possible:oWhen using a C100MMS-x cable, you can use a second C100MMS-x cable to daisy chain to an SCB100 accessory board, OR you could use a C50FF-x cable to daisy chain to a 50-pin accessory board.

oWhen using a C100HD50-x cable, you can use a C50FF-x cable to daisy chain to a 50-pin accessory
board.

Configuring the 6K-SSR-RACK08 board
The 6K-SSR-RACK08 has jumpers that you set to configure the power source and relay logic control. Factoryconfigured default settings are listed in Table 2-6.
Table 2-6. Default hardware configuration
Board label

Description

Default setting

POWER IN
JP1

Sets the power source to one of three options: +5BD: +5V power from the connected 6000 Series control board.
 +5PC: power from the computer’s internal power supply.
 9V EXT: power from an AC adapter.
Jumper to set the I/O module type for SSR modules 1-4.
Jumper to set the I/O module type for SSR modules 5-8.
Jumper to set the I/O module parity for SSR modules 1-4.
Options are invert (active high) and non-invert (active low).
Jumper to set the I/O module parity for SSR modules 5-8.
Options are invert (active high) and non-invert (active low).

+5BD

JP2
JP3
JP5
JP6

outputoutputNON-INVERT (active low)
NON-INVERT (active low)

Figure 2-4 shows the location of the jumpers and power connectors on the 6K-SSR-RACK08.

2-8

6K-SSR-RACK08 User's Guide

Installing the 6K-SSR-RACK08

Figure 2-4. 6K-SSR-RACK08 board jumpers and external power connectors

Power source
Set the power source with jumper JP1 (labeled POWER IN). You can power the 6K-SSR-RACK08 using one of
the following sources.


the 6000 Series control board’s internal +5 V power supply (default)
your computer’s +5 V power supply, or external +5 V supply (power cable supplied)
AC adapter (supplied)Figure 2-5 shows the POWER IN jumper configured for each power source.

9V EXT+5BD+5PC

Internal powerfrom the MCC board

External power fromthe computer’s power supply

External power froman AC adapter

Figure 2-5. Power source jumper JP1 configurations

External +5 V PC power connections
You can power the 6K-SSR-RACK08 with your computer’s +5 V power supply using the C-PCPOWER-10
cord that was shipped with the board. Each end of the cable has a keyed MOLEX type connector. To connect toyour computer’s internal power connectors, do the following:
1.

Turn off power to the computer and remove the cover.

2.

Connect one end of the C-PCPOWER10 power cord to one of your computer’s internal power connectors.
If necessary, refer to your computer's user manual for the location of the power supply connectors.
2-9

6K-SSR-RACK08 User's Guide

3.

Installing the 6K-SSR-RACK08

Run the power cable out the back of the computer through an expansion slot or other opening, and replace
the cover on the computer.

Caution!

Be careful not to pinch the cable when you replace the cover — if this cable is cut, the resulting
short circuit can damage the computer.

4.

Connect the other end of the power cord to the connector labeled PC POWER connector (P14) on the 6KSSR-RACK08. Refer to Figure 2-4 for the location of this connector.

5.

Set the POWER IN jumper (JP1) for +5PC.

External +9V AC adapter power connection
You can power the 6K-SSR-RACK08 using the AC power adapter (part number CB-PWR-9). This adapter
provides 9 volts, 1 amp DC power, 110 VAC power. To connect the AC adapter, do the following:
1.

Turn off power to the computer.

2.

Connect the CB-PWR-9 cable to the 9V EXTERNAL POWER connector (P13) on the 6K-SSR-RACK08.
Refer to Figure 2-4 for the location of this connector.

3.

Plug the AC adapter into a power outlet.

4.

Set the POWER IN jumper (JP1) for 9V EXT.

I/O type
Set the type for each module group for either input or output (default). You cannot mix the I/O configuration
within a group.
Jumper JP2 sets the I/O type for SSR modules 1-4
Jumper JP3 set the I/O type for SSR modules 5-8.

Refer to Figure 2-4 for the location of these jumpers. Figure 2-6 shows the SSR modules 1-4 configured for
input (jumper position 2-3), and SSR modules 5-8 configured for output (jumper position 1-2).

OUT

OUT

1
2
3

JP2

1
2
3

JP3

IN

IN

SSR modules 1-4
configured for input

SSR modules 5-8
configured for output

Figure 2-6. I/O module type jumpers

Relay logic
Set the relay logic for each module group to be either inverted (active high) or non-inverted (active low,default). Jumper JP5 sets the logic for SSR modules 1 to 4. Jumper JP6 sets the relay logic for SSR modules 5
to 8. The logic settings are labeled INVERT and NON-INVERT on the board.
When you set the logic jumper to NON-INVERT, the SSR modules activate when the digital output signal
is low.
When you set the logic jumper to INVERT, the SSR modules activate when the digital output signal is
high.

2-10

6K-SSR-RACK08 User's Guide

Installing the 6K-SSR-RACK08

Figure 2-7 shows SSR modules 1-4 set to invert logic (active high), and modules 5-8 set to non-invert logic(active low).

JP5

JP6

INVERT

NON-INVERTINVERT

NON-INVERT

3 2 1

3 2 1

SSR modules 1-4
configured for active high

SSR modules 5-8
configured for active low

Figure 2-7. Relay logic jumpers JP5 and JP6

The I/O type and the relay logic settings control what happens when a module activates. For example, a module
group configured for SSR input with non-inverted logic outputs a low logic level when the module is ON. A
module group configured for SSR output with non-inverted logic requires a low logic level to turn the output
module ON.

2-11Chapter 3

Functional Details
6K-SSR-RACK08 functions are illustrated in the block diagram shown here.

5V
regulator+5V

SSR modules 1-4

Input power-+
2

Channel 2 SSR

Relay
driver

4

I/O
type

-+
3

Channel 3 SSR

OUT

4

-+
4

Relay control bits
DIO (0:3)-+
1

Channel 1 SSR

Channel 4 SSR
IN

Load

4

Relay
driver-+
6

Relay control bits
DIO (4:7)-+
5

Channel 5 SSR

Channel 6 SSR-+
7Channel 7 SSR
OUT

I/O
type

4
IN

Channel 8 SSR-+
8100-pin I/O connector

PCI-DAS6000 Series or PCI-DAC6700 Series board+5V

SSR modules 5-8

Screw terminal block

Figure 3-1. 6K-SSR-RACK08 functional block diagram

3-1

6K-SSR-RACK08 User's Guide

Functional Details

Components
Major components on the 6K-SSR-RACK08 are shown in Figure 3-2.

Figure 3-2. 6K-SSR-RACK08 component locations

Table 3-1 describes each component.
Table 3-1. 6K-SSR-RACK08 component descriptions
Callout

Board
label

Description

1

P12, P11

100-pin interface connectors. Connect to the 6000 Series control board or accessory board.

2
3
4
5
6

JP3
JP6
JP2
JP5
JP1

7
8
9
10

P13
P14
A1 to A8
±1 to ±811

D1 to D8

12
13

RN1
P9, P10I/O type jumper for SSR modules 1-4.
Relay logic jumper for SSR modules 1-4.
I/O type jumper for SSR modules 5-8.
Relay logic jumper for SSR modules 5-8.
Main +5 V power jumper to select the power source — internal power from the 6000 Series
control board, external power from the computer, or external power from the AC adapter.
External power connector. Connect to the AC adapter.
External power connector. Connect to the computer's +5 V supply.
Mounting locations for SSR modules <1:8>.
Relay screw terminals for field wiring connections. Two screw terminals are dedicated toeach SSR module.
LEDs that indicate the status of the SSR modules. Refer to page 3-4 for a summary of the
LED states.
Pull-up / pull-down resistor location.
50-pin interface connectors. Connect to the 6000 Series control board or accessory board.

3-2

6K-SSR-RACK08 User's Guide

Functional Details

Compatible SSR modules
The 6K-SSR-RACK08 board has locations for eight solid state relay modules that you can install in your 6KSSR-RACK08 board. The SSR modules use a standard color scheme so you can quickly identify what module
type is installed. Mounting screw threads are provided for you to easily install the SSR modules.
MCC offers SSR modules that are compatible with the 6K-SSR-RACK08: SSR-IAC-05
 SSR-IAC-05A
 SSR-IDC-05
 SSR-IDC-05NP
 SSR-OAC-05

 SSR-OAC-05A
 SSR-ODC-05
 SSR-ODC-05A
 SSR-ODC-05R
Details on these SSR modules are available from our web site atwww.mccdaq.com/cbicatalog/directory.asp?dept_id=246&top_id=25.

Screw terminal connections
Use the board's screw terminals to connect external devices to the SSR modules. Two terminals are dedicated toeach module (one positive and one negative terminal).
Caution!

Before connecting wires to the screw terminals, turn off the power to the 6K-SSR-RACK08, andmake sure that the signal wires do not contain live voltages.

Wire gauge
Use 12 AWG to 22 AWG wire to connect field devices. Properly insulate the wires to avoid any short circuit tothe other channels, ground, or other points on the board.

Signal level control and power up conditions
To ensure a known relay state when the 6K-SSR-RACK08 is disconnected from the 6000 Series control board,or when the digital lines on the 6000 Series control board are in high impedance (or input) mode, install a 2.2 K
single inline package (SIP) resistor network.
In a 2.2 K SIP, one side of each resistor connects to a single common point and brought out to a pin. The
common point is marked with a line at one end of the resistor network. A typical SIP resistor network is shownin Figure 3-3.
2.2 K SIP

Common
line

LO

HI

I/O lines

Figure 3-3. SIP resistor location

The 6K-SSR-RACK08 has a location for installing a 2.2 K SIP resistor network (see Figure 3-4). Wheninstalling pull-up and pull-down resistor SIP packs, we recommend using a 2.2 K, eight-resistor SIP (MCC partnumber SP-K2.29C).

3-3

6K-SSR-RACK08 User's Guide

Functional Details

Figure 3-4. SIP resistor location

To pull up the digital line for a particular port, install the resistor with the common line at the HI end. To pull
down a digital line, install the resistor with the common line at the LO end. The SIP establishes a high or low
logic level at each digital I/O line when disconnected from the 6000 Series control board or when the digital
lines on the 6000 Series control board are in high impedance (or input) mode.
When the 6K-SSR-RACK08 is disconnected from the 6000 Series control board or when the digital lines on the6000 Series control board are in high impedance (or input) mode, the solid state relays are undriven. The stateof the solid state relays when not driven is dependent on the invert/non-invert logic jumper setting and the pullup/pull-down resistor configuration. Table 3-1 summarizes the undriven state of the SSR output modules foreach configuration.
Table 3-1. SSR output module power up and reset conditions
I/O type

Logic jumper setting

Pull-up / down
installed

Power up / reset
state

outputinvert (logic high)outputnon-invert (logic low)none
pull-uppull-downnone
pull-uppull-downhigh state = onhigh state = onhigh state = off
low state = on
low state = off
low state = on

LED states
The LEDs indicate the on/off status of the SSR modules. The LED illuminates when an output module is active(turned on) or when an input module detects an input voltage (logic high).

3-4

6K-SSR-RACK08 User's Guide

Functional Details

6K-SSR-RACK08 assembly diagram

3-5Chapter 4

Specifications
Typical for 25 °C unless otherwise specified.
Specifications in italic text are guaranteed by design.

Power consumption
Table 1. Power consumption specifications
All modules off
All modules on
All modules off
All modules on

5 V PC auxiliary power / PCI bus
power
External (9 V to 15 V) unregulated
supply200 mA typical, 230 mA max.
260 mA typical, 320 mA max.
210 mA typical, 240 mA max.
270 mA typical, 330 mA max.

Environmental
Table 2. Environmental specifications
Operating temperature range
Storage temperature range
Humidity

0 to 70 °C-40 to 100 °C
0 to 95% non-condensing

Mechanical
Table 3. Mechanical specifications
Card dimensions with modules(without standoffs)

248 mm (L) x 102 mm (W) x 41 mm (H)9.75" (L) x 4.0" (W) x 1.625" (H)Relay screw terminals
Table 4. Relay screw terminal specifications
Wire gauge range

12 AWG to 22 AWG
Table 5. Screw terminal pin out

Pin
1+
12+
23+
34+
45+
56+
67+
78+
8-

Signal Name
Module 1+
Module 1Module 2+
Module 2Module 3+
Module 3Module 4+
Module 4Module 5+
Module 5Module 6+
Module 6Module 7+
Module 7Module 8+
Module 8-4-1

6K-SSR-RACK08 User's Guide

Specifications

Power in jumper (JP1)
Table 6. JP1 specifications+5PC+9V EXT+5 BD (default)

Use cable C-PCPOWER-10
Use Adapter CB-PWR-9
Powered from 100-pin connector

I/O module type selection (JP2, JP3)
Table 7. JP2 and JP3 specifications
Modules 1-4
Modules 5-8
Pull-up/pull-down on digital I/O lines

Selectable via JP2 as either input modules or output (default) modules. Do not
mix input and output modules within this bank of four.
Selectable via JP3 as either input modules or output (default) modules. Do not
mix input and output modules within this bank of four.
Configurable at RN1 with 2.2 K resistor network. Not populated by default.

I/O module polarity selection (JP5, JP6)
Table 8. JP5 and JP6 specifications
Inverted (active high) or non-inverted (active low, default), selectable via JP5.
Inverted (active high) or non-inverted (active low, default), selectable via JP6.

Modules 1-4
Modules 5-8

Bypass resistors
Table 9. Bypass resistor specifications
Resistors R1 – R8
(Normally 0 Ω)

Transceiver bypass resistors for bit-wise I/O configuration. NOT populated by default. Bypass
resistors are mutually exclusive of 74LS245 and 74LS640 transceivers.

Compatible products
Table 10. Compatible product specifications
Analog input boards

Analog output boards

 PCI-DAS6013
 PCI-DAS6014
 PCI-DAS6030
 PCI-DAS6031
 PCI-DAS6032
 PCI-DAS6033
 PCI-DAS6034
 PCI-DAS6035
 PCI-DAS6036
 PCI-DAS6052
 PCI-DAS6023
 PCI-DAS6025
 PCI-DAS6040
 PCI-DAS6070
 PCI-DAS6071
 PCI-DAC6702
 PCI-DAC6703

Note 1: The 6K-SSR-RACK08 requires external power (for all products above) when used with the

C100HD50 (pins 51-100) ribbon cable.
4-2

6K-SSR-RACK08 User's Guide

Specifications

Main connectors and pin out
User connector P11
Table 11. Connector P11 specifications
Connector typeCompatible cables

Shielded SCSI 100 D-type
C100MMS-x, shielded round cable. x = 1, 2 or 3 meters
Table 12. P11 pin out

Pin
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18
19
20
21
22
23
24
25
26
27
28
29
30
31
32
33
34
35
36
37
38
39
40
41
42
43
44
45
46
47
48
49
50

Signal name
P12 Pass Through 1
P12 Pass Through 2
P12 Pass Through 3
P12 Pass Through 4
P12 Pass Through 5
P12 Pass Through 6
P12 Pass Through 7
P12 Pass Through 8
P12 Pass Through 9
P12 Pass Through 10
P12 Pass Through 11
P12 Pass Through 12
P12 Pass Through 13
P12 Pass Through 14
P12 Pass Through 15
P12 Pass Through 16
P12 Pass Through 17
P12 Pass Through 18
P12 Pass Through 19
P12 Pass Through 20
P12 Pass Through 21
P12 Pass Through 22
P12 Pass Through 23
P12 Pass Through 24
P12 Pass Through 25
P12 Pass Through 26
P12 Pass Through 27
P12 Pass Through 28
P12 Pass Through 29
P12 Pass Through 30
P12 Pass Through 31
P12 Pass Through 32
P12 Pass Through 33
P12 Pass Through 34
P12 Pass Through 35
P12 Pass Through 36
P12 Pass Through 37
P12 Pass Through 38
PC +5VP12 Pass Through 40
P12 Pass Through 41
P12 Pass Through 42
P12 Pass Through 43
P12 Pass Through 44
P12 Pass Through 45
P12 Pass Through 46
P12 Pass Through 47
P12 Pass Through 48
P12 Pass Through 49
GND

Pin
51
52
53
54
55
56
57
58
59
60
61
62
63
64
65
66
67
68
69
70
71
72
73
74
75
76
77
78
79
80
81
82
83
84
85
86
87
88
89
90
91
92
93
94
95
96
97
98
99
100

4-3

Signal name
P12 Pass Through 51
P12 Pass Through 52
P12 Pass Through 53
P12 Pass Through 54
P12 Pass Through 55
P12 Pass Through 56
P12 Pass Through 57
P12 Pass Through 58
P12 Pass Through 59
P12 Pass Through 60
P12 Pass Through 61
P12 Pass Through 62
P12 Pass Through 63
P12 Pass Through 64
P12 Pass Through 65
P12 Pass Through 66
P12 Pass Through 67
P12 Pass Through 68
P12 Pass Through 69
P12 Pass Through 70
P12 Pass Through 71
P12 Pass Through 72
P12 Pass Through 73
P12 Pass Through 74
P12 Pass Through 75
P12 Pass Through 76
P12 Pass Through 77
P12 Pass Through 78
P12 Pass Through 79
P12 Pass Through 80
P12 Pass Through 81
P12 Pass Through 82
P12 Pass Through 83
P12 Pass Through 84
DIO0
DIO1
DIO2
DIO3
DIO4
DIO5
DIO6
DIO7
P12 Pass Through 93
P12 Pass Through 94
P12 Pass Through 95
P12 Pass Through 96
P12 Pass Through 97
P12 Pass Through 98
P12 Pass Through 99
GND

6K-SSR-RACK08 User's Guide

Specifications

User connector P12
Table 13. Connector P12 specifications
Connector typeCompatible cables

Shielded SCSI 100 D-type
C100MMS-x, shielded round cable. x = 1, 2 or 3 meters
Table 14. P12 pin out

Pin
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18
19
20
21
22
23
24
25
26
27
28
29
30
31
32
33
34
35
36
37
38
39
40
41
42
43
44
45
46
47
48
49
50

Signal name
P11 Pass Through 1
P11 Pass Through 2
P11 Pass Through 3
P11 Pass Through 4
P11 Pass Through 5
P11 Pass Through 6
P11 Pass Through 7
P11 Pass Through 8
P11 Pass Through 9
P11 Pass Through 10
P11 Pass Through 11
P11 Pass Through 12
P11 Pass Through 13
P11 Pass Through 14
P11 Pass Through 15
P11 Pass Through 16
P11 Pass Through 17
P11 Pass Through 18
P11 Pass Through 19
P11 Pass Through 20
P11 Pass Through 21
P11 Pass Through 22
P11 Pass Through 23
P11 Pass Through 24
P11 Pass Through 25
P11 Pass Through 26
P11 Pass Through 27
P11 Pass Through 28
P11 Pass Through 29
P11 Pass Through 30
P11 Pass Through 31
P11 Pass Through 32
P11 Pass Through 33
P11 Pass Through 34
P11 Pass Through 35
P11 Pass Through 36
P11 Pass Through 37
P11 Pass Through 38
PC +5VP11 Pass Through 40
P11 Pass Through 41
P11 Pass Through 42
P11 Pass Through 43
P11 Pass Through 44
P11 Pass Through 45
P11 Pass Through 46
P11 Pass Through 47
P11 Pass Through 48
P11 Pass Through 49
GND

Pin
51
52
53
54
55
56
57
58
59
60
61
62
63
64
65
66
67
68
69
70
71
72
73
74
75
76
77
78
79
80
81
82
83
84
85
86
87
88
89
90
91
92
93
94
95
96
97
98
99
100

4-4

Signal name
P11 Pass Through 51
P11 Pass Through 52
P11 Pass Through 53
P11 Pass Through 54
P11 Pass Through 55
P11 Pass Through 56
P11 Pass Through 57
P11 Pass Through 58
P11 Pass Through 59
P11 Pass Through 60
P11 Pass Through 61
P11 Pass Through 62
P11 Pass Through 63
P11 Pass Through 64
P11 Pass Through 65
P11 Pass Through 66
P11 Pass Through 67
P11 Pass Through 68
P11 Pass Through 69
P11 Pass Through 70
P11 Pass Through 71
P11 Pass Through 72
P11 Pass Through 73
P11 Pass Through 74
P11 Pass Through 75
P11 Pass Through 76
P11 Pass Through 77
P11 Pass Through 78
P11 Pass Through 79
P11 Pass Through 80
P11 Pass Through 81
P11 Pass Through 82
P11 Pass Through 83
P11 Pass Through 84
DIO0
DIO1
DIO2
DIO3
DIO4
DIO5
DIO6
DIO7
P11 Pass Through 93
P11 Pass Through 94
P11 Pass Through 95
P11 Pass Through 96
P11 Pass Through 97
P11 Pass Through 98
P11 Pass Through 99
GND

6K-SSR-RACK08 User's Guide

Specifications

User connector P9
Table 15. Connector P9 specifications
Connector typeCompatible cables

Unshielded 50-pin ribbon connector - male
C100HD50-x, C50FF-x, unshielded ribbon cable. x = 3 or 6 feet.
Table 16. P9 pin out

Pin
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18
19
20
21
22
23
24
25

Signal name
P10 Pass Through 51
P10 Pass Through 52
P10 Pass Through 53
P10 Pass Through 54
P10 Pass Through 55
P10 Pass Through 56
P10 Pass Through 57
P10 Pass Through 58
P10 Pass Through 59
P10 Pass Through 60
P10 Pass Through 61
P10 Pass Through 62
P10 Pass Through 63
P10 Pass Through 64
P10 Pass Through 65
P10 Pass Through 66
P10 Pass Through 67
P10 Pass Through 68
P10 Pass Through 69
P10 Pass Through 70
P10 Pass Through 71
P10 Pass Through 72
P10 Pass Through 73
P10 Pass Through 74
P10 Pass Through 75

Pin
26
27
28
29
30
31
32
33
34
35
36
37
38
39
40
41
42
43
44
45
46
47
48
49
50

4-5

Signal name
P10 Pass Through 76
P10 Pass Through 77
P10 Pass Through 78
P10 Pass Through 79
P10 Pass Through 80
P10 Pass Through 81
P10 Pass Through 82
P10 Pass Through 83
P10 Pass Through 84
DIO0
DIO1
DIO2
DIO3
DIO4
DIO5
DIO6
DIO7
P10 Pass Through 93
P10 Pass Through 94
P10 Pass Through 95
P10 Pass Through 96
P10 Pass Through 97
P10 Pass Through 98
P10 Pass Through 99
GND

6K-SSR-RACK08 User's Guide

Specifications

User connector P10
Table 17. Connector P10 specifications
Connector typeCompatible cables

Unshielded 50-pin ribbon connector - male
C100HD50-x, C50FF-x, unshielded ribbon cable. x = 3 or 6 feet
Table 18. P10 pin out

Pin
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18
19
20
21
22
23
24
25

Signal name
P9 Pass Through 51
P9 Pass Through 52
P9 Pass Through 53
P9 Pass Through 54
P9 Pass Through 55
P9 Pass Through 56
P9 Pass Through 57
P9 Pass Through 58
P9 Pass Through 59
P9 Pass Through 60
P9 Pass Through 61
P9 Pass Through 62
P9 Pass Through 63
P9 Pass Through 64
P9 Pass Through 65
P9 Pass Through 66
P9 Pass Through 67
P9 Pass Through 68
P9 Pass Through 69
P9 Pass Through 70
P9 Pass Through 71
P9 Pass Through 72
P9 Pass Through 73
P9 Pass Through 74
P9 Pass Through 75

Pin
26
27
28
29
30
31
32
33
34
35
36
37
38
39
40
41
42
43
44
45
46
47
48
49
50

4-6

Signal name
P9 Pass Through 76
P9 Pass Through 77
P9 Pass Through 78
P9 Pass Through 79
P9 Pass Through 80
P9 Pass Through 81
P9 Pass Through 82
P9 Pass Through 83
P9 Pass Through 84
DIO0
DIO1
DIO2
DIO3
DIO4
DIO5
DIO6
DIO7
P9 Pass Through 93
P9 Pass Through 94
P9 Pass Through 95
P9 Pass Through 96
P9 Pass Through 97
P9 Pass Through 98
P9 Pass Through 99
GND

Declaration of Conformity
Manufacturer:
Address:

Category:

Measurement Computing Corporation
10 Commerce Way
Suite 1008
Norton, MA 02766
USAElectrical equipment for measurement, control and laboratory use.

Measurement Computing Corporation declares under sole responsibility that the product

6K-SSR-RACK08
to which this declaration relates is in conformity with the relevant provisions of the following standards or otherdocuments:
EU EMC Directive 89/336/EEC: Electromagnetic Compatibility, EN 61326 (1997) Amendment 1 (1998)
Emissions: Group 1, Class A
EN 55011 (1990)/CISPR 11: Radiated and Conducted emissions.

Immunity: EN61326, Annex A








IEC 1000-4-2 (1995): Electrostatic Discharge immunity, Criteria C.
IEC 1000-4-3 (1995): Radiated Electromagnetic Field immunity Criteria A.
IEC 1000-4-4 (1995): Electric Fast Transient Burst immunity Criteria B.
IEC 1000-4-5 (1995): Surge immunity Criteria A.
IEC 1000-4-6 (1996): Radio Frequency Common Mode immunity Criteria A.
IEC 1000-4-8 (1994): Magnetic Field immunity Criteria A.
IEC 1000-4-11 (1994): Voltage Dip and Interrupt immunity Criteria A.

Declaration of Conformity based on tests conducted by Chomerics Test Services, Woburn, MA 01801, USA in
April, 2005. Test records are outlined in Chomerics Test Report #EMI3931.04.
We hereby declare that the equipment specified conforms to the above Directives and Standards.

Carl Haapaoja, Director of Quality Assurance

Measurement Computing Corporation
10 Commerce Way
Suite 1008
Norton, Massachusetts 02766(508) 946-5100
Fax: (508) 946-9500
E-mail: info@mccdaq.com
www.mccdaq.com