Journal of Low Temperature Physics, Vol. 140, Nos. 5/6, September 2005 (© 2005)
DOI: 10.1007/s10909-005-7322-5

Trapping Fermionic 40 K and Bosonic 87 Rb on a Chip
S. Aubin,1 M. H. T. Extavour,1 S. Myrskog,1 L. J. LeBlanc,1 J. Estève,2
S. Singh,1 P. Scrutton,1 D. McKay,1 R. McKenzie,1 I. D. Leroux,1A. Stummer,1 and J. H. Thywissen1
1 McLennan Physical Laboratories, University of Toronto, Toronto, Ontario, M5S 1A7 Canada
2 Laboratoire de Photonique et de Nanostructures, UPR 20 du CNRS, 91460 Marcoussis,

FranceE-mail: joseph.thywissen@utoronto.ca; http://www.physics.utoronto.ca/∼jhtgroup/(Received February 8, 2005)

We demonstrate the loading of a Bose–Fermi mixture into a microfabricated magnetic trap. In a single-chamber vacuum system, laser-cooled atoms aretransported to the surface of a substrate on which gold wires have been microfabricated. The magnetic ﬁeld minimum formed near these current-carrying wires is used to conﬁne up to 6 × 10 4 neutral 40K atoms. In addition,
we can simultaneously load 2 × 10 5 87Rb atoms, demonstrating the conﬁnement of two distinct elements with such a trap. In a sequence optimized
for 87Rb alone, we observe up to 1 × 10 7 trapped atoms. We describe indetail the experimental apparatus, and discuss prospects for evaporative cooling towards quantum degeneracy in both species.
KEY WORDS: Ultracold atoms; magnetic trapping; microfabrication;
Bose–Einstein condensation; Fermi degenerate gas; micro-electromagnet.

1. INTRODUCTION
Since the observation of Bose–Einstein condensation1 and Fermi
degenerate gases,2 atomic gases have been used to study quantum degenerate many-body systems. Experiments have demonstrated exquisite control over the trapping environment, temperature, density, and interaction
strength of the constituents. In contrast to solid state and liquid systems,
quantum degenerate gases are dilute and without impurities, making themmore easily compared to theoretical models. Furthermore, observations
can be made with spatial imaging or with momentum spectroscopy, both
3770022-2291/05/0900-0377/0 © 2005 Springer Science+Business Media, Inc.

378

S. Aubin et al.

of which can be done rapidly enough to observe the dynamics of the system. These systems can be tailored to access interesting many-body phenomena such as the Hubbard model,3 Anderson localization,4 and the
BEC-BCS crossover regime.5
Microfabrication of magnetic trapping elements brings new tools tothe study of ultra-cold quantum gases.6,7 Micro-electromagnets (µEMs)
can be used to build smaller traps, achieve stronger trap gradients and curvatures,8 and reduce the power dissipation of the trapping elements. In
addition, microfabrication techniques make integration with waveguides9
and other devices10,11 possible. The ability to trap atoms near a surface
has led to a range of recent studies focusing on interactions between
degenerate atom clouds and surfaces.12,13 To date, µEMs have only beenused to study thermal and degenerate Bose gases.
In this paper we discuss an apparatus built to study ultra-cold Fermi
gases in a µEM trap. The high oscillation frequency accessible in astrongly conﬁning µEM trap has several advantages for studies of quantum gases. First, an increased collision rate allows fast evaporative cooling,which simpliﬁes the realization of a dual species system, as described in
Section 2. Second, the high oscillation frequency allows quantum degeneracy with a smaller number of atoms. The Fermi temperature in a threedimensional harmonically trapped gas is
TF =

hω

kB

1

(6N ) 3 ,(1)

where ω is the mean oscillation frequency, and N is the atom number.
At ﬁxed T /TF , N is proportional to ω−3 , so higher oscillation frequencies
make possible the study of quantum degenerate gases with smaller atom
numbers. Systems with small numbers of trapped fermions are reminiscent of quantum dots, where small numbers of electrons are captured inelectrostatic wells. Mesoscopic neutral Fermi gas samples are also expected
to exhibit rich physical phenomena. While Coulomb interactions dominate
electron mesoscopics, the relatively weaker interaction energies of neutral
Fermi systems facilitate the study of purely quantum-statistical effects suchas spatial shell structure.14
Another appealing feature of µEM traps is the variety of trap geometries which can be formed. Single traps with a high aspect ratio (>103 )could be used to study quasi-one-dimensional ensembles of fermions15 orof Bose–Fermi mixtures.16 Two or more traps could be connected throughquantum point contacts17 or tunneling barriers.
In the following sections, we describe our progress towards realizing
some of the experiments mentioned above using bosonic 87 Rb and fermionic 40 K. After a description of the apparatus in Section 2, we discussTrapping Fermionic 40 K and Bosonic 87 Rb on a Chip379

its performance to date in Section 3, which includes the ﬁrst demonstration, to our knowledge, of trapping fermions in a µEM trap. We discussprospects for evaporative cooling towards quantum degeneracy in both
species and our experimental outlook in Section 4.
2. EXPERIMENTAL APPARATUS
Experiments with quantum degenerate gases require trapping, cooling,
and imaging of ultra-cold atoms. In our case, trapping is accomplishedby capturing atoms from a background vapor in a Magneto-Optical Trap
(MOT), and then transferring them to a purely magnetic trap, both within
the same vacuum chamber. Cooling is accomplished in two steps: ﬁrst, bylaser cooling in the MOT and with optical molasses; second, by forced
evaporative cooling in the µEM magnetic trap. Imaging is either ﬂuorescent or absorptive, using resonant laser light and CCD cameras in bothcases.
Our system is vastly simpliﬁed by the use of a µEM trap. The
evaporative cooling times realized in such traps are typically a few seconds,6,13,18,19 short enough that a single vacuum chamber can be usedboth to collect atoms from the background vapor and to cool them todegeneracy with evaporative cooling. Systems with magnetic traps createdby standard coils require multiple chambers (or Zeeman slowers) to maintain an ultra-high vacuum. Additional lasers are then typically required tomove atoms between chambers (or slow them). In a dual-species experiment such as ours, additional lasers would be required for both species.
The single-chamber conﬁguration possible with a µEM trap avoids theabove complications.
The following subsections describe the essential components of theexperiment, sometimes focusing on technical choices that we believe would
be interesting to researchers in the ﬁeld. For a more general audience, thisdescription can serve as a speciﬁc example of a degenerate gas experimentthat is structurally similar to many other existing experiments. We also
refer the reader to review articles of experimental techniques.20
2.1. Vacuum System
The required quality of the vacuum environment of the trap is in general dictated by the desired lifetime of the trapped atoms. A collision witha background atom or molecule (thermalized with the walls of the chamber at room temperature) can eject an atom from the trap, since the energy
transferred is generally much higher than the trap depth (between 1 K
and 1 mK). For a 10 s 1/e lifetime for rubidium, for instance, the partial

380

S. Aubin et al.

pressure of hydrogen must be no higher than 3 × 10−9 mbar.21 A vacuum
environment also serves to thermally isolate the cold gases from ambient
temperature, since convective and conductive heat transfer are eliminatedby working in a vacuum.22
Our vacuum system consists of a Pyrex cell, 75 mm × 75 mm × 165 mm,
connected to an ion pump and a titanium sublimation pump through a
6 inch conﬂat stainless steel cube a turbo pump is connected to the system,
during pump-down and bake-out, but sealed off using a gate valve during
normal operation.23 The entire system is supported on an optical table insuch a way that the glass cell is closest to the table. Magnetic trap (seeSection 2.4) lifetimes as long as 9 s and MOT lifetimes as long as 24 s havebeen observed, suggesting that the pressure in the system is at or below
4 × 10−9 mbar.
2.2. Sources
Alkali atoms – both rubidium and potassium – are selectively released
into the vacuum chamber using dispenser sources.24 These devices use areaction between an alkali salt and a reducing agent to produce free alkali
atoms and a byproduct. The reaction is activated by resistive heating of
the nichrome envelope in which the reactants are housed.
Dispensers are commercially available25 for both rubidium and potassium in their natural isotopic abundance. While the natural abundance of
87 Rb (27.83%) is acceptably high, the natural abundance of 40 K is only
0.012%. We use a commercial rubidium dispenser but have constructed
a potassium dispenser using an isotope-enriched potassium salt. Our dispenser was modeled after that of Ref. 26, with the addition of a ﬁne nickel
mesh to contain the reactants. The device uses the following heat-enhanced
reaction to supply potassium:
2KCl + Ca → 2K + CaCl2 .
Using hot wire tests, we estimate that our isotope-enriched dispenser
can produce 100 µg of fermionic potassium. Given our typical 40 K MOT
size of 107 atoms (discussed in Section 3.2), it is reasonable to expect thedispenser lifetime to be on the order of years, as has been observed inother degenerate gas experiments.
Although some atoms dispensed by our sources are trapped in the
MOT, most are deposited on the walls of the glass cell or captured by
the vacuum pumps. We observe that 87 Rb atoms can be “recycled” fromthe glass walls using light-induced atom desorption (LIAD).27,28 Whereas
previous observations of this effect in cold atom experiments have usedTrapping Fermionic 40 K and Bosonic 87 Rb on a Chip

381

broadband white light sources,6,29 we observe wavelength dependence of
the 87 RB desorption effeciency.30
With the dispensers off, we measured the number of 87 Rb atoms
captured by the MOT as a function of the center wavelength of various weak (∼0.5 mW) LED sources of desorbing light. Wavelengths greaterthan 500 nm had no visible effect, whereas wavelengths in the 400 nm
range boosted the MOT loading rate by a factor of ﬁve. This is consistent with observations in parafﬁn-coated vapor cells,28 although it isunknown if the LIAD mechanism is identical. Encouraged by these preliminary tests, we built a 140 mW source at a center wavelength of 405 nm.
Desorption using this source allows us to load 109 87 Rb atoms into the
MOT without turning on the dispensers,31 which reduces the backgroundpressure and increases the trap lifetime. The atom number we capture with
LIAD in a UHV cell is 100 times larger than without LIAD, and over 10times larger than that reported in previous experiments,6,19 where higher
power white light sources were used. This comparison may not be an
accurate measure of the source efﬁciency, since our MOT uses much larger
trapping beams. Nonetheless, LIAD allows us to load large MOTs while
maintaining a better vacuum for longer magnetic trap lifetimes.
2.3. Laser System
Near-resonant lasers are used for laser cooling and trapping, optical pumping, and imaging. These applications require laser linewidths thatare small compared to the natural atomic linewidth ( ≈ 2π × 6 MHz forthe cycling transitions in 40 K and 87 Rb) and sufﬁcient power to saturate
the cooling transition in our MOT. These constraints are met by locking
low-power grating-stabilized diode lasers, injecting into free-running seedlasers, passing the light through single-mode optical ﬁbers, and where necessary, injecting into a tapered ampliﬁer as a second stage of ampliﬁcation,
as shown in Fig. 1. The ﬁber optics produce clean spatial modes for imaging beams, and decouple the alignment of the beams around the vacuum
system from other, less critical alignments.
Four master lasers are used: two for the 87 Rb D2 line at 780.246 nm,
and two for the 40 K D2 line at 766.702 nm, Of each pair, one laser istuned to the cycling transition (i.e., from the 5S1/2 F = 2 ground state to
the 5P3/2 F  = 3 excited state for 87 Rb, and from the 4S1/2 F = 9/2 groundstate to the 4P3/2 excited state manifold for 40 K) and the other laser tothe repumping transition (out of the F = 1 ground state for 87 Rb and outof the F = 7/2 ground state for 40 K). The ground state hyperﬁne splittingsare 6.835 and 1.286 GHz for 87 Rb and 40 K respectively, large enough that
separate master lasers are convenient.

382

S. Aubin et al.

These resonant wavelengths are an ideal combination for a twospecies experiment.32 They are close enough that they can share thesame mirrors, ﬁbers, coatings, and tapered ampliﬁer (whose bandwidthis 23 nm), but far enough apart that dichroic waveplates can selectively
manipulate the polarization of one wavelength without affecting the other,such that 100% combining efﬁciency can be achieved.
Each master is locked using saturated absorption spectroscopy. A
modulation transfer scheme is used, in which the frequency of the pump is
dithered (in our case, at 100 kHz) and the resultant amplitude modulationof the probe is used to lock to a peak maximum. The output of a lockcircuit is sent both to the piezo voltage and to the current of the master
laser, to achieve a lock bandwidth of over 5 kHz. Using this method, we
observe a short term stability of 300 kHz. Three of these master lasers (allbut the Rb repump) inject room-temperature free-running diode lasers toboost their power.
Figure 1 shows the mixing of various frequencies into the ﬁber optics
that transport them to the vacuum system. The majority of the laser power
is injected into the tapered ampliﬁer, producing 0.65 W of output power
after spatial ﬁltering, which is split into six beams for the MOT. A second
ﬁber is used for imaging, which is described in more detail in Section 3.1. A
third ﬁber is used for optical pumping into a speciﬁc magnetic sublevel tomaximize efﬁciency of loading from the MOT (which traps all states) into
the magnetic trap (which traps only speciﬁc sublevels). In all cases, dichroic
waveplates33 are used to produce a single polarization of light.

Fig. 1. A schematic showing the splitting and combining of the the various optical frequencies according to their function and desired path through the vacuum chamber. Polarizing beam splitter cubes, broadband waveplates (black), and dichroic wave plates (grey) are
shown. Single-mode optical ﬁbers send light to the vacuum system. One of these ﬁbers injects
the tapered ampliﬁer, which produces the MOT beams. The other ﬁber paths are aligned
directly into the pyrex vacuum cell. Additional frequency modiﬁcations by acousto-optic
modulators are not shown.

Trapping Fermionic 40 K and Bosonic 87 Rb on a Chip383

2.4. Magnetic Trap
Two pairs of circular coils are used to create the magnetic ﬁeld for the
MOT, for the ﬁrst magnetic trap, and for the movement of the magnetically trapped atoms from the ﬁrst magnetic trap (formed at the MOT) tothe chip trap 5 cm away. As shown in Fig. 2, all four coils are mounted
outside the glass cell. The inner pair of coils (the “MOT coils”) create
the linear quadrupole ﬁelds for the MOT and magnetic trap. Both MOT
coils consist of 100 turns of insulated, hollow-core34 copper wire, wound
to form a square cross-section with an inner (outer) diameter of 10 cm(18.4 cm). The coils are mounted with an inner separation of 8.4 cm. The
outer coils (the “Transfer coils”) carry equal and parallel currents to provide a nearly uniform ﬁeld in the z direction. They are made of the same
hollow-core copper wire, but have 49 turns each and inner (outer) diameters of 28 cm (36.4 cm). The Transfer coils are mounted with an innerseparation of 11.4 cm. Applying the uniform ﬁeld of the Transfer coils to
the quadrupole ﬁeld of the MOT coils shifts the location of the B = 0
trap center along the z axis (vertical). The MOT laser beams are aligned
so that they intersect at z = −2.5 cm from the center of the MOT coils.

Fig. 2. Glass cell, MOT coils. Transfer coils, atom chip, and MOT beams as seen along (a)the x axis, (b) the y axis; and (c) the z axis, Coils (shaded) are positioned around the glass
vacuum cell (solid lines). Diagonal (ŷ ± ẑ) MOT beams are shown (dashed) only in (a); horizontal (x̂) MOT beams are shown in both (b) and (c). The atom chip is seen face-on in (c)and indicated with a thick black line in (a) and (b).

384

S. Aubin et al.

Together, these coils create and move the trap center between the locationof the MOT and the surface of the chip, several centimeters away.
The advantage of this “off-center” MOT is power efﬁciency. A strong
quadrupole trap can only be formed within roughly half a coil radius of
the center of a pair of anti-Helmholtz coils. Had we formed the MOT at
the center of the coils and transported the atoms the same distance to thechip, the radius of the coils would have been doubled. Maintaining thesame magnetic ﬁeld gradient in these larger coils would require an orderof magnitude more electrical power.
As built, the power consumption for the primary coils is 0.8 kW at
60 A, which generates a gradient of 94 G/cm along the x̂-axis. Opticalaccess is sufﬁcient for six 5-cm-diameter MOT beams. Using the Transfer
coils, cold atoms have been transported over 5.5 cm.
2.5. Trap Switch
Several phases of the experiment require fast switching of the magnetic
trap coils. In order to load atoms from the MOT into the magnetic trap,
the coils must be turned off quickly to create a zero-ﬁeld condition for
optical molasses, and then turned on quickly to high current for efﬁcientloading of the magnetic trap. Fast turn-off is also necessary when imaging
the atoms in order to avoid heating, distortion of the cloud, and Zeeman
shifts of the atomic energy levels. The necessary time scale is determinedby the motion of atoms in the trap: ﬁelds should be switched in less timethan a single classical oscillation period. A chip trap (see Section 2.6) canhave oscillation frequencies above 1 kHz, requiring sub-ms switching times
for its bias ﬁelds. At 3 mH inductance and 60 A of current, voltages up to
1 kV are required to achieve sufﬁciently fast switching times.
Fig. 3a shows a schematic of the control circuit (dubbed “Mag-OMatic”) for rapidly switching large currents. The complete trap requires
two such switches: one for the MOT and one for the Transfer coil. These
are independent and nearly identical; only the MOT control circuit isshown. The difference between the two circuits is the role of the relay: theTransfer coils are hard-wired in series, but the relay determines the direction of current; the relay in the MOT circuit switches the coils between
Helmholtz and anti-Helmoltz conﬁgurations.
When the coils are operating in steady state, the IGBT (Insulated
Gate Bipolar junction Transistor) stack is on and the coil supplies are running in constant current mode. The current ﬂows through the blocking
diode, the coils, the IGBT stack, and a giant magneto-resistive (GMR)
current sensor. The sensor is used for quantitative monitoring only.

Trapping Fermionic 40 K and Bosonic 87 Rb on a Chip

385

Fig. 3. (a) Magnetic trap switch for the MOT coils. The primary switch is a stack of Insulated Gate Bipolar junction Transistors (IGBT’s). Turn-on is accelerated with a high voltage(HV) charge in a capacitor, whose discharge is triggered using a Silicon Controlled Rectiﬁer(SCR). Turn-off is accelerated with transient voltage suppressors (TVS’s). Current is sensed
using a giant magneto-resistive (GMR) current sensor. Arrows indicate the direction of controls and monitors; half arrows indicate coil currents. (b) Turn-off performance: current falls
from 60 A to zero in 150 µs. (c) Turn-on performance: rise of current from 0 to 26.5 A in
350 µs.

The coils are turned off by turning the IGBT stack off. The counterEMF (or “ﬂyback”) caused by the inductance of the coils produces a highvoltage spike which is clamped by the transient voltage suppressors (TVS)at about 940 V. The coil current is dissipated at 0.4 A/µs, as shown in
Fig. 3b.
There are two ways to turn on the coil current. If the IGBT stackalone is turned on, the current will rise asymptotically to the steady-statevalue in ∼20 ms. The other turn-on method (the “fast on”) works as follows. At any time before the fast turn-on is needed, the high voltage (HV)
supply is used to charge up the HV capacitor. Fast turn-on is triggeredby ﬁrst turning on the IGBT stack, and then triggering the HV SiliconControlled Rectiﬁer (SCR). In the few microseconds between these two
events, no signiﬁcant current ﬂows due to the coil inductance. Triggering
the SCR creates an LC parallel resonant circuit with the coil inductanceand the HV capacitor. During the ﬁrst quarter-cycle of the resonance,roughly 350 µs, the capacitor transfers its charge to the coil (see Fig. 3c).
The charge on the capacitor is chosen such that after this quarter-cycle thecoil is at its steady state voltage, and the blocking diode of the supplies
will conduct to provide the steady state current. The SCR turns off when
the current ﬂowing through it falls to zero.

386

S. Aubin et al.

2.6. Atom Chip
Although the external magnetic trap described in Section 2.4 can capture and hold the atoms, efﬁcient evaporative cooling requires a collision rate that is several hundred times greater than the loss rate from thetrap.35 In our single-chamber vapor cell, the loss rate is roughly 0.2 s−1 .
As a result, the initial collision rate must be greater than 102 s−1 , which(for the number and temperature typically captured by a MOT) requires amore tightly conﬁning trap than our external coils can provide.
A µEM trap (or “atom chip”) can provide much stronger conﬁnement – while using smaller currents – than a trap produced by conventional coils.8 To understand this, consider a cylindrical wire of diameter
D carrying current I . The ﬁeld gradient at a distance r > D/2 from thewire isB = −µ0 I,
2π r 2(2)where µ0 is the permittivity of free space. Decreasing the wire radiusdecreases the minimum r but also decreases the current capacity36 I as
D 3/2 , and thus the maximum gradient scales as D −1/2 . Reducing D frommillimeter-scale to micrometer-scale therefore improves the magnetic ﬁeld
gradient by several orders of magnitude. This higher gradient provides
tighter conﬁnement, which increases the collision rate and accelerates thethermalization necessary for evaporative cooling.
The atom chip we are using for this experiment is identical to the one
described in Ref. 18. Two types of traps are used: a “U-trap” and a “Ztrap”. These conﬁgurations, along with the external ﬁeld necessary to forma trap near the wire, are shown in Fig. 4.
The U-trap consists of a 420-µm-wide, 7-µm-high gold wire in the
shape of the letter ‘U’, as shown in Fig. 4a. The base of the U is 2.0 mm
long. When a bias ﬁeld is applied parallel to the plane of the U andperpendicular to its base, a three-dimensional quadrupole trap is formed
above the base of the wire. Using a current of 5.0 A and a bias ﬁeld of
20 G, a quadrupole trap is formed 430 µm from the surface, with a trap
depth of 1.3 mK, and a mean gradient of 200 G/cm.
The Z-trap consists of a 60-µm-wide, 7-µm-high gold wire in the
shape of the letter ‘Z’, as shown in Fig. 4b. The middle segment of the
Z is 2.85 mm long. When a bias ﬁeld is applied to this wire, a IoffePritchard-type trap (i.e., a magnetic trap whose minimum ﬁeld is nonzero)is formed above the surface. Using a current of 2.0 A and a bias ﬁeld of
40 G, the minimum is 100 µm from the surface, with an oscillation frequency of 6.5 kHz in the transverse direction and 13 Hz in the longitudinalTrapping Fermionic 40 K and Bosonic 87 Rb on a Chip

387

Fig. 4. Schematic diagrams of the U (upper) and Z (lower) wire trap conﬁgurations. The
arrows on the chip surfaces indicate the direction of DC current ﬂow; a uniform bias ﬁeld isapplied in the direction indicated. These two trapping wires are fabricated on the same chip,
but shown separately here for clarity.

direction. These calculations are for 40 K, and assume a 1 G “Ioffe ﬁeld”
applied in the ŷ direction, to increase the minimum ﬁeld of the trap.
2.7. Sequencing
The experiment operates in a cycle, starting with the collectionof hot atoms and progressively compressing and cooling the cloud tohigher phase space density. Coordination of each stage of the experiment is accomplished using a dedicated real-time system sequencer.37
The sequencer operates with an 80 MHz clock rate, and has 128 MB
of onboard memory, allowing for the generation of complex waveforms.
Twenty-four analog outputs and 32 digital outputs are updated as quickly
as once every 10 µs. These outputs are buffered (as described below) andsent to various parts of the experiment: shutters, frequency modulators,

388

S. Aubin et al.

Fig. 5. Schematic diagram of one channel of the analog buffer.

power supplies, cameras, and the magnetic switch. A standard desktop
computer programs the sequencer over an ethernet connection.
In order to protect the sequencer, drive low impedance loads, andprevent ground loops, buffers are used for all channels of the sequencer.
Galvanic isolation of the digital channels is accomplished by using a
GMR-based device.38 The schematic used for the analog buffer is shownin Fig. 5. Outputs from the buffer have “soft” grounding: the 10 between
the source ground and the buffer output return is low enough to carry any
load current yet high enough to prevent ground loops. A variation on adifferential input ampliﬁer corrects for the output return’s offset and noise.
2.8. Evaporative Cooling System
Laser cooling of atoms is restricted to low densities because lightmust be able to escape from the cloud to carry away entropy. In essence,
this constraint limits most laser cooling schemes to phase space densities
below 10−4 . Further cooling can be accomplished in a purely magnetic
trap using evaporative cooling, in which the hottest atoms are selectively
removed from the trap, each carrying away more than the average energyof a trapped atom. After rethermalization, the remaining ensemble has alower temperature. Atoms are selectively removed using an electromagnetic
ﬁeld that couples only to atoms located at higher potential energy surfaces
in the trap. Generally, the ﬁeld is in the radio frequency (RF) range. The
RF ﬁeld spin-ﬂips atoms to a magnetically anti-trapped state, and they areejected from the trapping volume.
While many quantum gas experiments use standard commercial function generators, such devices typically have dwell times and switching timesof 10 ms (or greater), rendering them unable to sweep quickly enough foran evaporation that may last only one second. Instead we generate theRF ﬁeld using Direct Digital Synthesis (DDS), which has sufﬁcient speed
and precision for our application, at a fraction of the cost of a completeTrapping Fermionic 40 K and Bosonic 87 Rb on a Chip

389

high-frequency function generator. DDS creates a sinusoidal waveform
point by point, which is then sent through a digital to analog converter.39
Using dividers implemented in digital logic, any frequency from µHz up
to the Nyquist frequency (146 MHz in our case) can be generated with 48bit resolution. Linear sweeps can be produced by specifying beginning andending frequency, the number of frequency steps, and step time. Since nophase-locked loop is involved, the sweep rate is limited only by the digitalclock cycle. Exponential sweeps are approximated piecewise by a series oflinear sweeps.
The frequency and amplitude of the DDS are programmed using
three digital lines from the sequencer. The RF output signal is ampliﬁed
and fed into an auxiliary wire on the chip, which acts as an “antenna”.
Since the antenna is less than a millimeter away from the atoms, little
power is required to perform RF evaporation.
2.9. Imaging
We obtain information about the atom cloud with ﬂuorescence andabsorption imaging. Fluorescence imaging is accomplished by observing
the light of a resonant laser beam scattered by an atom cloud. In absorption imaging, a resonant beam is passed through the cloud, and imaged
on a CCD camera. The atoms absorb a portion of the laser beam, leaving behind a shadow of the atom cloud. Since the optical density of thecloud is proportional to the three-dimensional density integrated along theline of observation, the shadow gives direct information about the spatial
distribution of the atom cloud.
At the chip, we also detect the cold atomic cloud by generating a onedimensional Magneto-Optical Trap (1D MOT). A retro-reﬂected probe
laser parallel to the chip surface is directed onto the magnetic ﬁeld minimum of the chip trap, where the atoms are located. When the probe laser
is red-detuned from resonance, the cold atomic cloud scatters light, while
being conﬁned by a magneto-optical forces in the direction of the probe
propagation. The longer interrogation time in the 1D MOT allows fora higher signal-to-noise detection than simple ﬂuorescence imaging. The
method is useful for measuring atom number, but it does not image thecloud, since the atoms are pushed towards the magnetic center of the 1D
MOT. (In fact, this technique is a useful way to determine the locationsof magnetic ﬁeld minima above the chip).
Four different imaging systems are used in the experiment: one looking from below along the ẑ-axis, two to image clouds at the chip from thex̂ and ŷ directions, and one to image the MOT and ﬁrst magnetic trap.
Each imaging system makes use of two lenses in order to form an image

390

S. Aubin et al.

of the cloud on the CCD of the camera. The two chip imaging systems
are built to observe small clouds, and use a matched set of achromats withnumerical apertures of 0.33 for the x̂-axis system and 0.25 for the ŷ-axis
system.
We use monochrome CCD cameras with 640 × 480 pixels, 7.4 µm ×
7.4 µm each, with a variable gain 10-bit intensity greyscale.40 The acquisition duration is electronically shuttered and can be varied in steps of
50 µs. Communication with imaging software over an IEEE1394 (“Firewire”) communication standard obviates the need for a dedicated imaging
board.
3. TRAPPING OF 87 RB AND 40 K ON A CHIP
Even though the experimental apparatus described in Section 2 is stillin the ﬁnal stages of construction and testing, we have successfully cooled
and trapped atoms on the chip. In the following paragraphs we describe(in Section 3.1) the sequence we use for loading cold 87 Rb, and (in Section
3.2) a recent result in which we have used a variation on the sequence tosimultaneously load 87 Rb and 40 K into a chip trap.
3.1. Experimental Sequence for 87 Rb
From the background vapor produced by a rubidium dispenser, wetrap and cool ∼109 87 Rb atoms in a magneto-optical trap (MOT). The
trapping beams each have an intensity of about 2.5 mW/cm2 , and are
detuned δ = −4.5 from the 5S1/2 F = 2 → 5P3/2 F = 3 cycling transition.
The repumper is 0.8% of the trapping intensity. The MOT coils provide
a gradient of 9 G/cm along the strong axis. We cool the atoms further
by rapidly turning off the current in the anti-Helmholtz coils and detuning the trapping beams to δ = −9 for 9 ms to create an optical molasses.
After this second cooling stage, we measure a temperature of T ≤ 30 µK
and a phase space density ρ ≈ 3 × 10−6 .
After the molasses stage, the atoms are recaptured in a purely magnetic trap with a vertical gradient of 21 G/cm. This magnetic ﬁeld gradient is strong enough to trap atoms in the F = 2, mF = 2 hyperﬁne groundstate against the force of gravity, while atoms in lower magnetic substatesremain untrapped. Once trapped, the atoms are compressed adiabaticallyfor transport to the chip with a vertical gradient of 47 G/cm (94 G/cm
on the strong horizontal axis of the trap). Typically, we load 15% of theatoms (1.5 × 108 atoms) into the quadrupole trap, and after compression,
they have a temperature of T ≈ 100µK and a phase space density ofρ ≈ 10−7 .

Trapping Fermionic 40 K and Bosonic 87 Rb on a Chip

391

The efﬁciency of the quadrupole trap loading is improved by theinsertion of a short optical pumping step immediately after the opticalmolasses stage. We turn on a uniform bias magnetic ﬁeld to provide aquantization axis for the atoms and then apply along this same axis a
200 µs laser pulse tuned to the 5S1/2 F = 2 → 5P3/2 F = 2 transition to optically pump the atoms. By using σ + polarized light, the atoms are pumped
into the F = 2, mF = 2 hyperﬁne ground state. This step typically improves
the loading efﬁciency by a factor of two, while leaving the size and temperature of the atomic cloud essentially unchanged.
We transport the atoms in the quadrupole trap up to the chip bychanging the current in the Transfer coils. With a magnetic ﬁeld gradient of 47 G/cm in the vertical direction, we move the atoms from the
MOT position to the chip by smoothly varying the vertical (+ẑ) bias magnetic ﬁeld (produced by the Transfer coils) from −84 to +121 G in about
600 ms. The atoms do not experience any signiﬁcant loss or heating during
the transport. Horizontal (x̂ and ŷ) bias magnetic ﬁelds provide transverse
positioning to align the cloud to the center of the µEM trap.
Once at the chip, the atoms are loaded into the U-wire trap by ﬁrst
increasing the current in the U-wire, and then simultaneously decompressing the quadrupole trap while increasing the transverse horizontal bias
magnetic ﬁeld for the chip trap. The loading process takes 300 ms. Afterloading the U-wire trap, we ﬁnd that the trap contains 1 × 107 atoms witha temperature of T = 200 µK and a phase space density of ρ ≈ 10−7 .
This entire sequence can be initiated at a lower vapor pressure byturning off the dispensers and using LIAD, as is described in Section 2.2.
Figure 6 shows images of light-desorbed 87 Rb in the MOT (Fig. 6a) andafter release from the chip trap (Fig. 6b and c).
3.2. Simultaneous Loading of 87 Rb and 40 K
Recently, we have adapted the above sequence, developed for 87 Rb, toinclude 40 K. We report two technical milestones: (1) trapping of a cold
fermionic atomic species in a chip trap, and (2) simultaneous trapping of
two elements, in this case a bosonic isotope and a fermionic isotope, in achip trap.
In order to load 40 K onto the chip, we make the following changes
to the sequence. We turn on the 40 K-enriched dispenser and potassium
lasers to load a MOT. Then, instead of using an optical molasses coolingstage, we compress the MOT for 100 ms with an anti-Helmholtz gradientof 94 G/cm. The trapping beams are then turned off and the atoms remaintrapped in the magnetic quadrupole ﬁeld (which is already on at full

392

S. Aubin et al.

Fig. 6. Images of 87 Rb (left column) and 40 K (right column) clouds in the magneto-opitcal
trap and-at the chip: (a) 87 Rb MOT; (b) 87 Rb after µEM trapping, imaged using a 1D MOT
at the chip; (c) Absorption image of 87 Rb after release from the µEM trap; (d) 40 K MOT;(e) 40 K after µEM trapping, imaged using a 1D MOT at the chip. More detail about atom
numbers and imaging techniques are given in the text. All images use the same spatial scale,
which is indicated.

strength). The rest of the sequence is as described in Section 3.1: we transport the cold atoms to the chip and transfer them into the µEM trap.
Using this sequence, 40 K is loaded onto the chip. Figure 6d shows a
40 K MOT and Fig. 6e shows a 40 K cloud that was released from the µEM
trap and imaged using the 1D-chip-MOT ﬂuorescence detection scheme
described in Section 2.9.
Turning on the LIAD source and 87 Rb lasers, we observe that both
species are loaded onto the chip, although 87 Rb is not loaded as efﬁciently
as in the original sequence. By selecting a probe beam at 780 or 767 nm,
we can detect 87 Rb or 40 K atoms, independently. We observe simultaneousTrapping Fermionic 40 K and Bosonic 87 Rb on a Chip

393

trapping of 2 × 105 87 Rb atoms and 6 × 104 40 K atoms in the U-wire
trap.
3.3. Discussion and Future Work
While the above sequences are successful, they are not optimal. The
transfer efﬁciency (∼10%) from the external quadrupole trap to the U-wireµEM trap is not as high as has been previously observed in similar transfer sequences.13 It is possible that we are limited by the trap depth of theµEM trap.
The sequence used to load 40 K is preliminary and has not been studied carefully. Unlike the sequence described in Section 3.1. the method
described in Section 3.2 to load the magnetic trap will trap more than one
mF level in each species. A spin mixture in the magnetic trap is susceptible
to spin exchange losses. The range of magnetic sublevels can be reducedwith an optical pumping step.
Future work includes the evaporative cooling of 87 Rb and sympathetic cooling of 40 K, both to quantum degeneracy. For quantum degeneracy, atoms must be loaded into the Z-wire trap, whose Ioffe-Pritchard ﬁeld
conﬁguration avoids spin-ﬂip losses. Improvements in loading the chip will
be necessary to achieve the correct starting condition for evaporative cooling to degeneracy.
Several additions to our apparatus are possible. Manipulation of theatoms with microwaves will enable the simultaneous evaporation of 40 K
and 87 Rb, which has been shown to improve efﬁciency in other dual species experiments.41 Far off-resonant laser light can be used to form a trap
using the induced electric dipole of the atom. Such an optical trap is useful for the study of multiple spin states, and for accessing Feshbach-Verhaar resonances.42 A next-generation microfabricated trap will be built to
reduce light scattering off the chip, and to address known problems withthe techniques used to fabricate our chip.18 Finally, a higher quality camera will improve our imaging sensitivity.
4. CONCLUSION
In conclusion, we have demonstrated the trapping of 40 K and of 87 Rb
on a chip. To our knowledge, this is the ﬁrst time fermions have been conﬁned by a µEM trap, and the ﬁrst time two elements have been simultaneously loaded into a µEM trap. The apparatus used to achieve this has
been described in detail, indicating some technical choices that may interest researchers in the ﬁeld.

394

S. Aubin et al.

The high aspect ratio of these traps may be useful for studies ofquasi-one-dimensional ensembles. In the simplest case, the aspect ratio iscontrolled by the ratio of the wire current to the bias ﬁeld: the closer
the trap is to the surface, the higher its aspect ratio. The 500:1 aspectratio of the Z-trap in the example given in Section 2.6. seems well suited
to quasi-one-dimensional experiments. A complication is the trap “roughness,” which has been investigated in Refs. 12, 18. This roughness may
spoil the use of some wire traps as simple elongated traps, but may also
allow for the study of quantum gases in potentials with frozen-in randomness.43
Note added in proof. Since the original submission of this paper, wehave been able to load simultaneously 2 × 10−7 87 Rb atoms with a phasespace density of 3 × 10−6 , along with 2 × 10−5 40 h atoms in the |F =9/2, mF = 9/2 > state into the z-wire trap.
ACKNOWLEDGMENTS
We would like to thank Hyun Youk and Ryan Bolen for early workon this project. We also thank Thorsten Schumm, Brian DeMarco, Vladan
Vuletić, Peter Krüger, Sebastian Hoferberth, Jörg Schmiedmayer, andJoszef Fortágh for stimulating and helpful conversations. S.A. and L.L.
acknowledge support from NSERC; M.H.T.E. acknowledges support from
OGS. This work was supported by CFI, OIT, NSERC, PRO, and Research
Corporation.
REFERENCES
1. M. H. Anderson, J. R. Ensher, M. R. Matthews, C. E. Wieman, and E. A. Cornell,
Science 269, 198 (1995); K. B. Davis, M.-O. Mewes, M. R. Andrews, N. J. van Druten,
D. S. Durfee, M. Kurn, and W. Ketterle, Phys. Rev. Lett. 75, 3969 (1995); C. C. Bradley,
C. A. Sackett, J. J. Tollett, and R. G. Hulet, Phys. Rev. Lett. 75, 1687 (1995); see also
C. A. Sackett, C. C. Bradley, M. Welling, and R. G. Hulet, Appl. Phys. B B65, 433(1997).
2. B. DeMarco and D. S. Jin, Science 285, 1703 (1999); A. G. Truscott, K. E. Strecker, W. I.
McAlexander, G. B. Partridge, and R. G. Hulet, Science 291,2570 (2001); F. Schreck,
L. Khaykovich, K. L. Corwin, G. Ferrari, T. Bourdel, J. Cubizolles, and C. Salomon,
Phys. Rev. Lett. 87, 080403 (2001); S. R. Granade, M. E. Gehm, K. M. O’Hara, and J. E.
Thomas, Phys. Rev. Lett. 88, 120405 (2002); Z. Hadzibabic, C. A. Stan, K. Diechmann,
S. Gupta, M. W. Zwierlein, A. Görlitz, and W. ketterrle, Phys. Rev. Lett. 88, 160401(2002); G. Roati, F. Riboli, G. Modugno, and M. Inguscio, Phys. Rev. Lett. 89, 150403(2002); M. Bartenstein, A. Altmeyer, S. Riedl, S. Jochim, C. Chin, J. Hecher Denschlag,
and R. Grimm, Phys. Rev. Lett. 92, 120401 (2004); M. Köhl, H. Moritz, T. Stöferle,
K. Günter, T. Esslinger Phys. Rev. Lett. 94, 080403 (2005); C. Silber, S. Guenther, C.
Marzok, B. Deh, Ph. W. Courteille, C. Zimmermann. cond-mat/0506217; C. Ospelkaus,
S. Ospelkaus, H. Sengstoch, K. Bongs cond-mat/0507219.

Trapping Fermionic 40 K and Bosonic 87 Rb on a Chip

395

3. W. Hofstetter, J. I. Cirac, P. Zoller, E. Demler, and M. D. Lukin, Phys. Rev. Lett. 89,
220407 (2002).
4. U. Gavish and Y. Castin, Phys. Rev.Lett. 95, 020401 (1995).
5. C. A. Regal, M. Greiner, and D. S. Jin, Phys. Rev. Lett. 92, 040403 (2004); M. Bartenstein, A. Altmeyer, S. Riedl, S. Jochim, C. Chin, J. Hecker Denschlag, and R. Grimm,
Phys. Rev. Lett. 92, 120401 (2004); T. Bourdel, L. Khaykovich, J. Cubizolles, J. Zhang,
F. Chevy, M. Teichmann, L. Tarruell, S. J. J. M. F. Kokkelmans, C. Salomon, Phys. Rev.
Lett. 93, 050401 (2004); M. W. Zwierlein, C. A. Stan, C. H. Schunck, S. M. F. Raupach,A. J. Kerman, W. Ketterle, Phys. Rev. Lett. 92, 120403 (2004).
6. W. Hänsel, P. Hommelhoff, T. W. Hänsch, and J. Reichel, Nature 413, 498 (2001).
7. H. Ott, J. Fortagh, G. Schlotterbeck, A. Grossmann, and C. Zimmermann, Phys. Rev.
Lett. 87, 230401 (2001).
8. J. D. Weinstein and K. G. Libbrecht, Phys. Rev. A, 52, 4004 (1995).
9. J. H. Thywissen, M. Olshanii, G. Zabow, M. Drndić, K. S. Johnson, R. M. Westervelt,
and M. Prentiss, Eur. Phys. J. D 7, 361 (1999); D. Müller, D. Z. Anderson, R. J. Grow, P.
D. D. Schwindt, and E. A. Cornell, Phys. Rev. Lett. 83, 5194 (1999); N. H. Dekker, C. S.
Lee, V. Lorent, J. H. Thywissen, S. P. Smith, M. Drndić, R. M. Westervelt, and M. Prentiss, Phys. Rev. Lett. 84, 1124 (2000).
10. W. Hänsel, J. Reichel, P. Hommelhoff, and T. W. Hänsch. Phys. Rev. Lett. 86, 608 (2001).
11. R. Folman, P. Krüger, D. Cassettari, B. Hessmo, T. Maier, and J. Schmiedmayer, Phys.
Rev. Lett. 84, 4749 (2000).
12. J. Fortágh, H. Ott, S. Kraft, A. Günther, and C. Zimmermann, Phys. Rev. A 66,
041604(R) (2002); M. P. A. Jones, C. J. Vale, D. Sahagun, B. V. Hall, and E. A. Hinds
Phys. Rev. Lett. 91, 080401 (2003); J. M. McGuirk, D. M. Harber, J. M. Obrecht, andE. A. Cornell Phys. Rev. A 69, 062905 (2004).
13. Y. Lin. I. Teper, C. Chin, and V. Vuletić Phys. Rev. Lett. 92, 050404 (2004).
14. J. Schneider and F. Wallis, Phys. Rev. A 57, 1253 (1998).
15. B. E. Granger and D. Blume, Phys. Rev. Lett. 92, 133202 (2004).
16. K. Das, Phys. Rev. Lett. 90, 170403 (2003); M. A. Cazalilla and A. F. Ho, Phys. Rev.
Lett. 91, 150403 (2003).
17. J. H. Thywissen, R. M. Westervelt, and M. Prentiss, Phys. Rev. Lett. 83, 3762 (1999).
18. J. Estève. C. Aussibal, T. Schumm, C. Figl, D. Mailly, I. Bouchoule, C. I. Westbrook, and
A. Aspect, Phys. Rev. A 70, 043629 (2004).
19. S. Du, M. B. Squires, Y. Imai, L. Czaia, R. A. Saravanan, V. Bright, J. Reichel, T. W.
Hänsch, and D. Z. Anderson, Phys. Rev. A 70, 053606 (2004).
20. See, for intance, W. Ketterle, D. S. Durfee, and D. M. Stamper-Kurn, in Bose–Einsteincondensation in atomic gases. Proceedings of the International School of Physics “Enrico
Fermi”, Course CXL, M. Inguscio, S. Stringari, and C. E. Wieman (eds.), IOS Press,
Amsterdam, (1999) pp. 67–176 (cond-mat/9904034). A more detailed description of one
BEC expriment is given in H. J. Lewandowski, D. M. Harber, D. L. Whitaker, and E. A.
Cornell, J. Low Temp. Phys. 132, 309 (2003).
21. J. E. Bjorkholm, Phys. Rev. A 38, 1599 (1988).
22. Radiative transfer is unaffected by the vacuum, but not signiﬁcant because the absorption frequencies of ground-state alkali atoms are far above the emission range of a blackbody at room temperature.
23. The ion pump is a Varian VacIon Plus 75 StarCell pump (75 1/s); the sublimation pumpuses a Varian 8” CF Cryopanel; the turbo pump used is a Pfeiffer Vacuum TMU071P;
the gate valve used is a VAT 10836-UE44-0005 DN63 2–1/2”.
24. C. Wieman, G. Flowers, and S. Gilbert, Am. J. Phys. 63, 317 (1995); J. Fortagh,
A. Grossmann, T. W. Hänsch, and C. Zimmermann, J. Appl. Phys. 84, 6499 (1998).
25. Alkali metal dispeners are available from SAES Getters.
26. B. DeMarco, H. Rohner, and D. S. Jin, Rev. Sei. Instr. 70, 1967 (1999).
27. A. Gozzini, F. Mango, J. H. Xu, G. Alzetta, F. Maccarrone, and R. A. Bernheim, Il
Nuovo Cimento 15, 709 (1993).

396

S. Aubin et al.

28. M. Meucci, E. Mariotti, P. Bicchi, C. Marinelli, and L. Moi, Europhys. Lett. 25, 639(1994).
29. B. P. Anderson and M. A. Kasevich, Phys. Rev. A 63, 023404 (2001).
30. LIAD is also effective for both 39 K and 40 K, but we have not studied this efﬁciency as afunction of wavelength.
31. Leaving the dispensers off for several weeks, we observe a slow decay in the atom number in the MOT, as if a diminishing amount of 87 Rb is accessible to desorption. Thisbehavior has also been observed by Dana Anderson and coworkers at JILA.
32. J. Goldwin, S. B. Papp, B. DeMarco, and D. S. Jin, Phys. Rev. A 65, 021402 (2002).
33. These dichroic waveplates act as a λ/2 waveplate for 780 nm and a λ waveplate for
767 nm. They were designed and custom built by OptiSource LLC.
34. The wire is hollow to allow the ﬂow of pressurized cooling water through the centre of
the wire. It was manufactured by Wolverine Fabricated Products and insulated by S&W
Wire.
35. W. Ketterle and N. J. van Druten, Adv. At. Mol. Opt. Phys. 37, 181 (1996).
36. S. Groth, P. Krüger, S. Wildermuth, R. Folman, T. Fernholz, D. Mahalu, I. Bar-Joseph,and J. Schmied-mayer, cond-mat/0404141 (2004).
37. We use an ADwin Pro sequencer, from Jäger Computergesteuerte Messtechnik GmbH.
38. Isolation for both inputs and outputs is done by the NVE 1L715-3 quad GMR-based
device. The Texas Instruments SN64BCT25244NT octal buffer gives a tolerant 50  output to the array of BNC connectors.
39. The DDS chip we use is the Analog Device’s AD9854, and comes on an evaluation
board costing roughly US $200. In our implementation, an on-board 15.360 MHz reference oscillator is multiplied up 19 times to give a 291.84 MHz sample frequency.
40. MicroPix cameras are used (model M640), with the window covering the CCD chip
removed by the manufacturer.
41. K. E. Strecker, G. B. Partridge, and R. G. Hulet, Phys. Rev. Lett. 91, 080406 (2003).
42. E. Tiesinga, B. J. Verhaar, and H. T. C. Stoof, Phys. Rev. A 47. 4114 (1993); S. Inouye.
M. R. Andrews, J. Stenger, H.-J. Miesner, D. M. Stamper-Kurn, and W. Ketterle, Nature
392, 151 (1998); J. L. Roberts, N. R. Claussen, J. P. Burke, Jr., Chris H. Greene, E. A.
Cornell, and C. E. Wieman, Phys. Rev. Lett. 81, 5109 (1998).
43. D. Wang, M. D. Lukin, and E. Demler, Phys. Rev. Lett. 92, 076802 (2004).