6220

Ethernet-Based Voltage Measurement Module
Features• 12 voltage inputs• 16-bit, 100 kHz per channel
sample rate
• ±10 V input range• Eight digital I/O• Simultaneous sampling• BNC connectors• Multiple trigger modes• Compact, portable design• Expandable using multiple
6000 Series modules
The 6220 with Encore software is a portable, voltage measurement solution

Software• Includes Encore interactive measurement software for Out-of-the-Box
setup, acquisition, display, logging,
analysis, and reporting• Advanced feature set with no programming required• Drag-and-drop functionality• Easy-to-use set-up wizards• Ability to drag data directly to Excel®
• Included data analysis tools• Playback mode
• Sophisticated test report capability• Incorporate multiple 6000 Series
devices into one application• Supported Windows® Operating
Systems: Windows 7 (32-bit or 64-bit),Windows Vista (32-bit or 64-bit),Windows XP SP2 (32-bit),Windows 2000 SP4

Overview
The 6000 Series combines accurate,
instrument class, Ethernet-based modules
with powerful, easy-to-use software. It isa major advancement in ease of use andfunctionality. 6000 Series modules feature
integrated signal conditioning and a modular
design that allows for system expansion.
Because they use an Ethernet connection,
modules can be connected directly to a PC
or used in remote configurations utilizing
multiple modules.
Each 6000 Series module includes Encore
interactive measurement software.
Encore couples ease-of-use with advanced
functionality including custom data
displays, powerful analysis, and detailed
reporting capabilities.

Measurement Computing

6000 Series Module Overview
6220

6222

6224

6230

6231

FeatureAnalog input typevoltagethermocouple

strain gage/bridge

voltagevoltage

Analog inputs

12

12

12

12

12

Digital I/O

8

8

8

8

8

Resolution

16 bit24 bit24 bit24 bit24 bitSample rate

50 kHz per channel

100 kHz per channel

2 Hz

50 kHz per channel

50 kHz per channel

Voltage input range±10 V±80 mV±25 mV/V

±10 V

±60 V

Connector type

BNC

screw-terminal

RJ50

screw-terminal

screw-terminal

Ch-to-ch isolation

—

—

—

Yes

Yes

Analog Inputs
The 6220 is equipped with 12 analog
input channels with a ±10 V range. Eachanalog input provides 16-bit resolution
with a maximum 100 kHz per channel
sample rate. All channels are sampled
simultaneously.

Digital I/O
The 6220 includes eight digital I/O lines.
All eight lines are accessible via a 9-pin
female DSUB connector located on the rear
panel of the unit. Each digital I/O bit canbe programmed individually to be either
an input or an output.

Encore provides real-time viewing andanalysis of data along with built-in test
report capabilities

Counters

Triggering

There are four 32-bit counters built into
each 6000 Series module. They are accessed
through the 9-pin female DSUB connector located on the rear of the unit. Eachcounter has a maximum 20 MHz input
frequency and can be used in counter or
encoder (A, B, and Z) modes.

A variety of trigger modes are supported by
the 6220. A wide selection of programmableanalog and digital trigger modes are available for starting an acquisition. All trigger
modes, along with the number of scans andthe sample rate for pre- and post-trigger
data, are software programmable prior to
the start of a scan sequence.

(508) 946-5100

1

info@mccdaq.com

mccdaq.com

6220

General Information
6220 System Examples
Notebook PC running
Encore software

Notebook PC running
Encore software

Ethernet

Ethernet network

Two 6220 voltage
measurement modules

6220 voltage
measurement module

System Power Connection

6220 Block Diagram

The 6220 and other 6000 Series modules
offer the flexibility to be powered either
directly from a 19 V to 30 VDC source, or
via the included TR-60U AC power adapter.

Power
Supply

CH 1
Diff
Amp

16-BitADC

Section 1
CommonControl
Logic

CH 2

Circuit is identical to channel 1 circuit.

CH 3

Circuit is identical to channel 1 circuit.

CH 4

Circuit is identical to channel 1 circuit.

CPU

10/100 MB
Ethernet

PHY
Isolator

The 6000 Series transfers acquired data to
the PC via 10/100BaseT Ethernet, allowing a continuous stream of data to becollected and stored in a PC’s memory or
hard drive. The most common and highestperformance connection is with dedicated,point-to-point Ethernet link between the
PC and the 6220. With an enterprise-wide
Ethernet network connection, any numberof 6000 Series modules can be connectedto the network.

Acquisition Controller

Ethernet Features

Data
Buffer
MemoryCalibration
MemoryMulti-Unit Synchronization

Measurement Computing(508) 946-5100

LEDs

CH 6
CH 7

The circuit layout is the same as the one
represented by the above block.

Isolator

CH 5

Digital I/O

CH 8

CH 9
CH 10
CH 11

The circuit layout is the same as the one
represented by the first block.

CH 12

2

info@mccdaq.com

Acquisition Controller

SYNC
Ports

Isolator

Multiple 6000 Series modules can besynchronized via the rear-panel SYNC
ports on each unit. After connecting
each module to an Ethernet port, simply
connect multiple 6000 Series modules
together using SYNC cables (CA-74-1).
Encore software provides the capability ofsetting one of the 6000 Series modules as
the master and the others as slaves. The
sampled data phase relationship amongchannels between multiple devices isdependent on the “Channel SYNC Skew”
specification for each device.

Chassis#6-32

mccdaq.com

6220

Specifications and Ordering Information
Specifications

The following specifications are typical for the temperature range -40 to 50 ˚Cunless otherwise noted.

Input Characteristics
Number of Analog Channels: 12
ADC Resolution: 16 bits
Type of ADC: Successive approximation register (SAR)
Input Range: ±10.0 V nominal; overflow will occur if | Vin | ≥ (10.2 V to 10.6 V)
Sampling Mode:  Simultaneous
Data Rates (fs):  1 S/s to 100 kS/s (16 selectable rates)
Multiple Device, Channel Sync Skew:  10 µs
Single Device, Channel-to-Channel Matching (Calibrated):  100 ns typical
Overvoltage protection ±30 V

Accuracy
Percent of Reading(Gain Error)

Offset Error

Calibrated, max (-40 to 50 ˚C)

0.2%±8.5 mV

Calibrated, typ (25 ˚C, ±5 ˚C)

0.02%±1.4 mV

Measurement Conditions

Encoder Sources: There are 3 encoder sources (A, B, and Z) that be assigned toany digital pin x
Counter Source: Internal clock, timer 1, timer 2, and digital pin x; one source canbe used in multiple counters
Counter Mode Options: Totalize, clear on read, rollover, stop at the top, increment,
decrement, rising edge, falling edge
Counter Gate Options: Unused, internal clock, timer 1, timer 2, and digital pin
x; one gate can be used in multiple counters

Stability
Gain Drift: 10 ppm/˚C
Offset Drift: 60 µV/˚C
CMRR (fin=60 Hz): -73 dB min
Input Bandwidth (-3 dB): 420 kHz min
Input Impedance
Resistance Between any Two AI-Terminals: 200 kOhm
Input Bias Current: 10 nA
Input Noise
RMS: 1.2 LSB
Peak-to-Peak: 7 LSB
Crosstalk: -80 dB
Settling Time [to 2 LSBs]
10 V Step: 25 µs
20 V Step: 35 µs
No Missing Codes: 15 bits guaranteed
DNL: ±2 LSB max
INL: ±6 LSB max

Power
Power Consumption: 5.5 W typ, 6 W max
Power Jack: Barrel type, 5.5 mm O.D.; 2.1 mm I.D.

Physical Characteristics
Weight: 1.36 Kg (3 lb)
Dimensions: 276.9 mm W x 169.8 mm D x 30.5 mm H (10.9” x 6.685” x 1.2”)

Environmental
The MCC 6220 is intended for indoor use only but may be used outdoors if installed
in a suitable enclosure.
Operating Temperature: -40 to 50 ˚C
Storage Temperature: -40 to 75 ˚C
Ingress Protection: IP 40
Operating Humidity: 10 to 90% RH, noncondensing
Storage Humidity: 5 to 95% RH, noncondensing
Maximum Altitude: 2,000 m (6,562 ft.)
Pollution Degree: 2

Digital I/O Lines
Channels: 8 digital I/O, programmable as a single port, or as individual lines
Power-Up Mode: Inputs pulled low
Connector: DB-9 female
Programmable Input Scanning Modes
Asynchronous: Under program control at any time relative to analog scanning
Synchronous: Data captured synchronously with the analog channels
Input Levels
Low: 0 to +0.8 V
High: +2.0 V to +5.0 V
Input Voltage Range without Damage: -0.6 V to +5.6 V max
Input Pull-Down Resistor: 10 kOhm
Output Voltage Range: 0 to +3 V (may be externally pulled up to +5.6 V without damage)
Output Resistance: 40 Ohm
Output Levels: (See 6000 Series Digital Output graph above)
Sampling: 1 MHz max continuous
Output Timing: Outputs are always written asynchronously

Calibration
Calibration Interval: 1 year; contact factory for information regarding calibration service.
Note: The above calibration information pertains to hardware calibration, notto be confused with “user” or “software” calibration. When performing a “user”
calibration via Encore (or other software) keep in mind that sample rate affects
both gain and offset of the hardware, and therefore any “user” calibration
should be performed at the same sample rate that is intended for measurements.

Counters
Channels: Up to 4 independent
Resolution: 32-bit
Input Frequency: 20 MHz max
Input Characteristics: 10 kOhm pulldown
Trigger Level: TTL
Minimum Pulse Width: 25 ns high, 25 ns low
Programmable Modes: Counter, encoder
Encoder Resolution: x1 (default), x2, and x4

Measurement Computing(508) 946-5100

3

info@mccdaq.com

mccdaq.com

6220

Ordering Information
Ordering Information

Description
12-channel, 16-bit resolution, Ethernet-based voltage measurement
module, with included Out-of-the-Box Encore software

Part No.
6220

Accessories & Cables
Universal power supply, 24 VDC @ 0.8 A (max); requires additional cable,
CA-1 (US) or CA-216 (European)
TR-60U
Required cable for use with TR-60U; 120 V US version
CA-1
Required cable for use with TR-60U; 220 V European version
CA-216
1
RJ12 shielded cable, 6 conductor, SYNC, 0.3 m (1 ft.)
CA-74-1
CA-192-7C
Ethenet crossover cables, 2.133 m (7 ft.) 2, 3
Ethernet patch cable, 0.457 m (1.5 ft.)2
CA-242
CA-242-7
Ethernet patch cable, 2.133 m (7 ft.)2
Stacking plate kit
190658A-01
Handle kit
HA-210-5-BK

1 Up to nine units can be synchronized. The total combined length of the SYNC
cables is not to exceed 2.438 m (8 ft.).
2 Ethernet cable length must be <3 m (9.8 ft.) in order for the system to be CE
Compliant.
3 Ethernet crossover cables should only be used for direct network connections.
In particular, attempting to connect a device to a Hub using a crossover cable
may prevent that network link from functioning. Some modern routers havebecome an exception by including logic to detect the crossover cable and allow
the network link to function.

Measurement Computing(508) 946-5100

4

info@mccdaq.com

mccdaq.com

6222

Ethernet-Based Thermocouple Input Module
Features• 12 thermocouple inputs• 24-bit ADC• ±80 mV input range• Simultaneous sampling• Eight digital I/O• Screw-terminal connectors• Multiple trigger modes• Ethernet interface to PC• Expandable using multiple 6000 Series
modules

Software• Includes Encore interactive measurement software for Out-of-the-Box
setup, acquisition, display, logging,
analysis, and reporting• Advanced feature set with no programming required• Drag-and-drop functionality• Easy-to-use set-up wizards• Ability to drag data directly to Excel®
• Included data analysis tools• Playback mode
• Sophisticated test report capability• Incorporate multiple 6000 Series
devices into one application• Supported Windows® Operating
Systems: Windows 7 (32-bit or 64-bit),Windows Vista (32-bit or 64-bit),Windows XP SP2 (32-bit),Windows 2000 SP4

Overview
The 6222 is a 12-channel thermocouple
input module. It features 24-bit ADCs forhighly accurate temperature measurements.
The 6222 is part of the 6000 Series
of instrument class, Ethernet-based
data acquisition modules. Modules
include powerful and easy-to-use Encore
Interactive Measurement Software.
Encore couples ease-of-use with advanced
functionality including custom data
displays, powerful analysis, and detailed
reporting capabilities. 6000 Series modules
feature integrated signal conditioning
and a modular design that allows for
system expansion. Because they use an
Ethernet connection, modules can beconnected directly to a PC or used in remoteconfigurations utilizing multiple modules.

Measurement Computing

The 6222 with Encore software is a portable, thermocouple measurement solution

6000 Series Module Overview
6220

6222

6224

6230

6231

FeatureAnalog input typevoltagethermocouple

strain gage/bridge

voltagevoltage

Analog inputs

12

12

12

12

12

Digital I/O

8

8

8

8

8

Resolution

16 bit24 bit24 bit24 bit24 bitSample rate

50 kHz per channel

100 kHz per channel

2 Hz

50 kHz per channel

50 kHz per channel

Voltage input range±10 V±80 mV±25 mV/V

±10 V

±60 V

Connector type

BNC

screw-terminal

RJ50

screw-terminal

screw-terminal

Ch-to-ch isolation

—

—

—

Yes

Yes

Analog Inputs
The 6222 is equipped with 12 thermocouple
input channels. Inputs are also equippedwith a ±80 mV range. Each analog input
provides 24-bit resolution with a 2 S/ssample rate per channel. All inputs are
sampled simultaneously.

Digital I/O
The 6222 includes eight digital I/O lines.
All eight lines are accessible via a 9-pin
female DSUB connector located on the rear
panel of the unit. Each digital I/O bit canbe programmed individually to be either
an input or an output.

With Encore software you can display
temperature and other channel data using
meters, gauges, tanks and graphs

Counters

Triggering

There are four 32-bit counters built into
each 6000 Series module. They are accessed
through the 9-pin female DSUB connector located on the rear of the unit. Eachcounter has a maximum 20 MHz input
frequency and can be used in counter or
encoder (A, B, and Z) modes.

A variety of trigger modes are supported by
the 6222. A wide selection of programmableanalog and digital trigger modes areavailable for starting an acquisition. All
trigger modes, along with the number ofscans and the sample rate for pre- and posttrigger data, are software programmableprior to the start of a scan sequence.

(508) 946-5100

1

info@mccdaq.com

mccdaq.com

6222

General Information
6222 System Examples
Notebook PC running
Encore software

Notebook PC running
Encore software

Ethernet

Ethernet network

6222 thermocouple
input module

Two 6222 thermocouple
input modules

System Power Connection

6222 Block Diagram

The 6222 and other 6000 Series modules
offer the flexibility to be powered either
directly from a 19 V to 30 VDC source, or
via the included TR-60U AC power adapter.

Ethernet Features

Removable
Screw-Terminal
Connector
COM

COM
No Connection

Internal Circuit (Channels 1-4) — See Note
Channel 1
Circuit—

CH1 TCCH2 TC
CH3 TCCH4 TC

CPU

+
—

CH2
CH3
CH4

+
—
+

MUX

24-BitADC

Control
Logic

10/100 MB
Ethernet

PHY

Isolator

The 6000 Series transfers acquired data to
the PC via 10/100BaseT Ethernet, allowing
a continuous stream of data to be collected
and stored in a PC’s memory or hard
drive. The most common and highestperformance connection is with dedicated,point-to-point Ethernet link between the
PC and the 6222. With an enterprise-wide
Ethernet network connection, any numberof 6000 Series modules can be connectedto the network.

Acquisition Controller

Power
SupplyData
Buffer
MemoryT—

Calibration
MemoryCJC+

Multi-Unit Synchronization

Measurement Computing(508) 946-5100

LEDs

Isolator

Internal Circuit (Channels 5-8) — See Note

Digital I/OInternal Circuit (Channels 9-12) — See Note

Acquisition Controller

SYNC
Ports

Isolator

Multiple 6000 Series modules can besynchronized via the rear-panel SYNC
ports on each unit. After connecting
each module to an Ethernet port, simply
connect multiple 6000 Series modules
together using SYNC cables (CA-74-1).
Encore software provides the capability ofsetting one of the 6000 Series modules as
the master and the others as slaves. The
sampled data phase relationship amongchannels between multiple devices isdependent on the “Channel SYNC Skew”
specification for each device.

Chassis#6-32

Note: Circuits for channel group 5 through 8 and group 9 through 12
look like that of the first group; for which only one channel is illustrated.
Each four-channel circuit has one 24-bit ADC and one CJC.

2

info@mccdaq.com

mccdaq.com

6222

Specifications
Specifications

The following specifications are typical for the temperature range -40 to 50 ˚Cunless otherwise noted.

Input Characteristics
Number of Channels: 12 thermocouple channels; 3 internal cold-junction compensation channels
ADC Resolution: 24 bits
Type of ADC: Delta-Sigma
Voltage Measurement Range: ±80 mV
Common-Mode Range
Channel-to-COM: ±1.5 V
Common-to-Earth Ground: ±250 V
Common-Mode Rejection Ratio (0 to 60 Hz)
Channel-to-COM: 95 dB
Common-to-Earth Ground: >170 dB
Temperature Measurement Ranges: Works over ranges defined by NIST for
J, K, R, S, T, N, E, and B thermocouple types
Cold-Junction Compensation Accuracy1
0 to 50 ˚C : 0.6 ˚C (1.1 ˚F) typ, 1.3 ˚C (2.3 ˚F) max-40 to 50 ˚C : 1.7 ˚C (3.1 ˚F) max
Data Rate (fs): 2 S/s
Input Bandwidth (-3 dB): 15 Hz
Noise Rejection: 85 dB min at 50/60 Hz
Overvoltage Protection: ±30 V between any input and common
Differential Input Impedance: 20 MOhm
Input Current: 50 nA
Input Noise: 1µ Vrms
Gain Error1: 0.05% max at 25 ˚C, 0.06% typ at -40 to 50 ˚C, 0.1% max at -40 to 50 ˚COffset Error1: 15 µV typ, 20 µV max
Gain Error from Source Impedance: 0.05 ppm per Ohm source impedance due
to input impedance
Offset Error from Source Impedance: 0.05 µV typ, 0.07 µV max per Ohm source
impedance due to input current

Counters

Power Consumption: 4.1 W max
Power Jack: Barrel type, 5.5 mm O.D.; 2.1 mm I.D.

Channels: Up to 4 independent
Resolution: 32-bit
Input Frequency: 20 MHz max
Input Characteristics: 10 kOhm pulldown
Trigger Level: TTL
Minimum Pulse Width: 25 ns high, 25 ns low
Programmable Modes: Counter, encoder
Encoder Resolution: x1 (default), x2, and x4
Encoder Sources: There are 3 encoder sources (A, B, and Z) that be assigned toany digital pin x
Counter Source: Internal clock, timer 1, timer 2, and digital pin x; one source canbe used in multiple counters
Counter Mode Options: Totalize, clear on read, rollover, stop at the top, increment,
decrement, rising edge, falling edge
Counter Gate Options: Unused, internal clock, timer 1, timer 2, and digital pin
x; one gate can be used in multiple counters

Physical Characteristics

SafetyPower

Weight: 1.22 Kg (2.7 lb)
Dimensions: 276.9 mm W x 169.8 mm D x 30.5 mm H (10.9” W x 6.685” D x 1.2”H)
Removable Screw-Terminal Blocks: 3 blocks, 4 channels per block, 10 terminals
per blockScrew-Terminal Wiring: 12 to 24 AWG wire with 10mm (0.39 in.) section of
insulation stripped from connection end
Torque for Screw Terminals: 0.5 to 0.6 N • m (4.4 to 5.3 lb • in.)Ferrules: 0.25 mm to 2.5 mm (optional, not supplied)

Safety Voltages
Only connect voltages that are within these limits.
Channel-to-COM: ±30 V max

Isolation Voltages
Channel-to-Channel
Within each Four-Channel Internal Module: No isolation between channels
Between each Four-Channel Internal Module: 250 V isolation

Digital I/O

Note: There is no isolation between channels 1, 2, 3, and 4; no isolation between
channels 5, 6, 7, and 8; and no isolation between channels 9, 10, 11, and 12.

Channels: 8 Digital I/O, programmable as a single port, or as individual lines:
Power-Up Mode: Inputs pulled low
Connector: DB-9 female (figure)
Programmable Input Scanning Modes
Asynchronous: Under program control at any time relative to analog scanning.
Synchronous: Data captured synchronously with the analog channels.
Input Levels
Low: 0 to +0.8 V
High: +2.0 V to +5.0 V
Input Voltage Range without Damage: -0.6 V to +5.6 V max
Input Pull-Down Resistor: 10 kOhm
Output Voltage Range: 0 to +3.0 V(may be externally pulled up to 5.6 V without damage)
Output Resistance: 40 Ohm
Output Levels: (See 6000 Series Digital Output graph above)
Sampling: 1 MHz max continuous
Output Timing: Outputs are always written asynchronously

Channel-to-Earth Ground Continuous: 250 Vrms, Measurement Category II
Note: Measurement Category II is for measurements performed on circuits directly
connected to the electrical distribution system. This category refers to local-level
electrical distribution, such as that provided by a standard wall outlet (for example,
115 V for U.S. or 230 V for Europe). Do not connect to signals or use for measurements within Measurement Categories III or IV.

1 Accuracy specifications are only valid when using provided connectors andproperly installed backshells.

Measurement Computing(508) 946-5100

3

info@mccdaq.com

mccdaq.com

6222

Specifications & Ordering Information
Environmental
The 6222 is intended for indoor use only but may be used outdoors if installed ina suitable enclosure.
Operating Temperature: -40 to 50 ˚C
Storage Temperature: -40 to 85 ˚C
Ingress Protection: IP 40
Operating Humidity: 10 to 90% RH, noncondensing
Storage Humidity: 5 to 95% RH, noncondensing
Maximum Altitude: 2,000 m (6,562 ft.)
Pollution Degree: 2

Calibration
Calibration Interval: 1 year; contact factory for information regarding calibration service.
Note: The above calibration information pertains to hardware calibration, notto be confused with “user” or “software” calibration. When performing a “user”
calibration via Encore (or other software) keep in mind that sample rate affects
both gain and offset of the hardware, and therefore any “user” calibration
should be performed at the same sample rate that is intended for measurements.

Ordering Information

Description
12-channel, 24-bit ADC, Ethernet-based thermocouple input
module, with included Out-of-the-Box Encore software

Part No.
6222

Accessories & Cables
Universal power supply, 24 VDC @ 0.8 A (max); requires additional cable,
CA-1 (US) or CA-216 (European)
TR-60U
Required cable for use with TR-60U; 120 V US version
CA-1
Required cable for use with TR-60U; 220 V European version
CA-216
RJ12 shielded cable, 6 conductor, SYNC, 0.3 m (1 ft.)1
CA-74-1
Ethenet crossover cables, 2.133 m (7 ft.) 2, 3
CA-192-7C
CA-242Ethernet patch cable, 0.457 m (1.5 ft.)2
Ethernet patch cable, 2.133 m (7 ft.)2
CA-242-7
Backshell connector kit
CN-271
Stacking plate kit
190658A-01
Handle kit
HA-210-5-BK

CN-271 connectors and backshells are included with the 6222 andprovide internal screw-terminal connectors for wiring thermocouples

1 Up to nine units can be synchronized. The total combined length of the SYNC
cables is not to exceed 2.438 m (8 ft.).
2 Ethernet cable length must be <3 m (9.8 ft.) in order for the system to be CE
Compliant.
3 An Ethernet crossover cable must be used when connecting a 6000 Series device
directly to the host PC.

Measurement Computing(508) 946-5100

4

info@mccdaq.com

mccdaq.com

6224

Ethernet-Based Strain Measurement Module
Features• 12 strain gage inputs• 24-bit/delta-sigma ADC per channel• 50 kHz per channel sample rate
• User programmable excitation• Simultaneous sampling• Remote sensing• Shunt calibration• Eight digital I/O• Multiple trigger modes• Compact, portable design• Expandable using multiple
6000 Series modules

Software

The 6224 with Encore software is a portable, strain gage measurement solution• Includes Encore interactive measurement software for Out-of-the-Box
setup, acquisition, display, logging,
analysis, and reporting• Advanced feature set with no programming required• True drag-and-drop functionality• Easy-to-use set-up wizards• Drag data directly to Excel®
• Included data analysis tools• Playback mode
• Sophisticated test report capability• Incorporate multiple 6000 Series
devices into one application• Supported Windows® Operating
Systems: Windows 7 (32-bit or 64-bit),Windows Vista (32-bit or 64-bit),Windows XP SP2 (32-bit),Windows 2000 SP4

Overview
The 6000 Series combines accurate, instrument class, Ethernet-based modules withpowerful, easy-to-use software. It is a majoradvancement in ease of use and functionality. 6000 Series modules feature integrated
signal conditioning and a modular designthat allows for system expansion. Because
they use an Ethernet connection, modules
can be connected directly to a PC or usedin remote configurations utilizing multiple
modules. Modules are offered for voltage or
direct strain inputs.
Each 6000 Series module includes Encore
interactive measurement software.
Encore couples ease-of-use with advanced
functionality including custom data
displays, powerful analysis, and detailed
reporting capabilities.
Measurement Computing

6000 Series Module Overview
6220

6222

6224

6230

6231

FeatureAnalog input typevoltagethermocouple

strain gage/bridge

voltagevoltage

Analog inputs

12

12

12

12

12

Digital I/O

8

8

8

8

8

Resolution

16 bit24 bit24 bit24 bit24 bitSample rate

50 kHz per channel

100 kHz per channel

2 Hz

50 kHz per channel

50 kHz per channel

Voltage input range±10 V±80 mV±25 mV/V

±10 V

±60 V

Connector type

BNC

screw-terminal

RJ50

screw-terminal

screw-terminal

Ch-to-ch isolation

—

—

—

Yes

Yes

Direct Strain Gage Inputs
The 6224 is equipped with 12 analog input
channels capable of measuring full-, half-,or quarter-bridge sensors. Each of the
12 analog channels has its own 24-bit ADC
and input amplifier allowing all 12 channels
to be sampled simultaneously. The 6224 also
includes anti-aliasing filters. All strain input
signals are attached via 12 RJ50 connectors
mounted on the front of the module.
The 6224 features direct strain gage inputs.
Half or full-bridge sensors are supported.
Quarter-bridge sensors are also supported
using an external resistor. The 6224 is also
100% software programmable. Excitation,
auto-zero, and shunt calibration are eachadjusted in software without relying on
manually adjusted trimpots which cancause increased error and drift. Excitation
is provided by the 6224 and can besoftware selected for 2.5 V, 3.3 V, 5 V,(508) 946-5100

1

Encore provides real-time viewing andanalysis of data along with built-in test
report capabilitiesor 10 V. For excitation values beyond thoselisted, an external excitation source may be
used. Connectors are provided on the frontof the unit for excitation source wiring.

info@mccdaq.com

mccdaq.com

6224

General Information
6224 Block Diagram
External
Excitation**To CH 1 thru 4

Isolation

To CH 5 thru 6

CH 1

To CH 7 thru 12+RS

Acquisition Controller

x0.25
Vref

xc

—E

+E
xc

Power
Supply

—RS
+In

x10

In

24-Bit
ADC—In
Shunt Cal
TEDS

CH 2

Circuit is identical to channel 1 circuit.

CH 3

Circuit is identical to channel 1 circuit.

CH 4

Circuit is identical to channel 1 circuit.

Data
Buffer
MemoryCalibration
MemoryExt. Exc**

CH 6
CH 7

LEDs
Channels 5 thru 8 and anExternal Excitation Source Connector

Isolator

CH 5

10/100 MB
Ethernet

PHY
Isolator

Control
Logic

CPU

The circuit layout is the same as the one
represented by the above block.

Digital I/O

CH 8
SYNC
Ports

CH 10
CH 11

Channels 9 thru 12 and anExternal Excitation Source Connector

Isolator

CH 9

The circuit layout is the same as the one
represented by the first block.

CH 12

Acquisition Controller

Ext. Exc**

Chassis#6-32** There are three External Excitation Voltage Source Connectors.
The first serves channels 1 thru 4.
The second serves channels 5 thru 8.
The third serves channels 9 thru 12.

Shunt Calibration

Remote Sensing

Shunt calibration enables each channel tobe put into a known imbalance condition
to set or verify channel calibration. Shuntcal allows a full scale gain to be set without
physically loading the bridge to capacity.

Remote sensing automatically and continuously corrects for voltage drop errors
in excitation leads.

Measurement Computing(508) 946-5100

2

info@mccdaq.com

mccdaq.com

6224

General Information
6224 System Examples
Notebook PC running
Encore software

Notebook PC running
Encore software

Ethernet

Ethernet network

Two 6224 strain
measurement modules

6224 strain
measurement module

Triggering

The CN-268 is an RJ50 to 12-pin screw
terminal break-out connector module. The
CN-269 and CN-270 offer internal bridge
completion resistors for quarter-bridge
configurations.

A variety of trigger modes are supported by the 6224. A wide
selection of programmable analog and digital trigger modes areavailable for starting an acquisition. All trigger modes, alongwith the number of scans and the sample rate for pre- and posttrigger data, are software programmable prior to the start of ascan sequence.

System Power Connection

Signal Connections
The 12 strain gage inputs on the 6224 are accessed via 12 RJ50
front panel connectors. 12 user specified connector modules areincluded with the 6224.

The 6224 and other 6000 Series modules offer the flexibility to
be powered either directly from a 19 V to 30 VDC source, or via
the included TR-60U AC power adapter.

Full- and Half-Bridge Configurations
The 6224 supports full- and half-bridge configurations directly.
Users can connect to the RJ50 front panel connectors or use theCN-268 12-pin screw terminal break-out connector module.

Ethernet Features
The 6000 Series transfers acquired data to the PC via 10/100BaseT
Ethernet, allowing a continuous stream of data to be collected
and stored in a PC’s memory or hard drive. The most commonand highest-performance connection is with dedicated, pointto-point Ethernet link between the PC and the 6224. With anenterprise-wide Ethernet network connection, any number of6000 Series modules can be connected to the network.

Quarter-Bridge Configurations
For quarter-bridge configurations the CN-269 and CN-270 feature
120 Ohm and 350 Ohm internal bridge completion resistors
respectively. The CN-269 and CN-270 feature an RJ50 female
connector one end and screw terminals on the other. One module
is required for each channel.

Multi-Unit Synchronization
Multiple 6000 Series modules can be synchronized via the rearpanel SYNC ports on each unit. After connecting each module toan Ethernet port, simply connect multiple 6000 Series modules
together using SYNC cables (CA-74-1). Encore software provides thecapability of setting one of the 6000 Series modules as the masterand the others as slaves. The sampled data phase relationshipamong channels between multiple devices is dependent on the“Channel SYNC Skew” specification for each device.

Digital I/O
The 6224 includes eight digital I/O lines. All eight lines are accessible via a 9-pin female DSUB connector located on the rear panelof the unit. Each digital I/O bit can be programmed individually
to be either an input or an output.

Counters
There are four 32-bit counters built into each 6000 Series module.
They are accessed through the 9-pin female DSUB connector
located on the rear of the unit. Each counter has a maximum
20 MHz input frequency and can be used in counter or encoder(A, B, and Z) modes.
Measurement Computing(508) 946-5100

3

info@mccdaq.com

mccdaq.com

6224

Specifications
Specifications

Input Noise
Excitation
Voltage

Density(nV/Vrms per
1 Hz )

Total(50 kS/s)(µV/Vrms)

0 to 1 kHz(nV/Vrms)

Input Characteristics

2.5 V

8

1.3

250

Number of Analog Channels: 12
Bridge Completion
Full and Half: Internal
Quarter: External
ADC Resolution: 24 bits
Type of ADC: Delta-sigma (with analog pre-filtering)
Sampling Mode: Simultaneous
Data Rates (fs): (50 kS/s)/n , n = 1, 2, ... 31.
Multiple Device, Channel SYNC Skew: 1 sample period
Single Device, Channel-to-Channel Matching (Calibrated): 350 nS (max)Master Timebase (Internal)
Frequency: 12.8 MHz
Accuracy: ±100 ppm max
Nominal Full-Scale Range: ±25 mV/V
Scaling Coefficient: 2.9802 nV/V per LSB
Overvoltage Protection Between any Two Terminals: ±30 V

3.3 V

6

1.0

190

5.0 V

4

0.6

130

10.0 V2

0.3

65

The following specifications are typical for the temperature range -40 to 50 ˚Cunless otherwise noted.

Spurious-Free Dynamic Range (SFDR): 106 dB, (1kHz, –60 dBFS)
Total Harmonic Distortion (Fundamental @ –20 dBFS)
1 kHz: -100 dB
8 kHz: -90 dB
Excitation Noise: 0.1 mV/Vrms
Crosstalk1 kHz: -110 dB
10 kHz: -100 dB
Shunt Calibration
Resistance: 100 kOhm
Resistor Accuracy
25 ˚C: ±110 Ohm-40 to 50 ˚C: ±200 Ohm
Excitation
Internal Voltage: 2.5 V, 3.3 V, 5.0 V, 10.0 V
Internal Power: 450 mW max
External Voltage: 2 V to 10 V

Accuracy
Error*

Percent of Reading

OffsetCalibrated max (-40 to 50 ˚C)

0.20%

0.0625 mV/V

Calibrated typ (25 ˚C, ±15 ˚C)

0.05%

0.0125 mV/V

Digital I/O* Excluding offset null or shunt calibration.

Channels: 8 digital I/O, programmable as single port, or as individual lines
Power-Up Mode: Inputs pulled low
Connector: DB-9 female
Programmable Input Scanning Modes
Asynchronous: Under program control at any time relative to analog scanning
Synchronous: Data captured synchronously with the analog channels
Input Levels
Low: 0 to +0.8 V
High: +2.0 V to +5.0 V
Input Voltage Range without Damage: -0.6 V to +5.6 V max
Input Pull-Down Resistor: 10 kOhm
Output Voltage Range: 0 to +3 V, (may be externally pulled up to 5.6 V
without damage)
Output Resistance: 40 Ohm
Output Levels: (See 6000 Series Digital Output graph below)
Sampling: 1 MHz max continuous
Output Timing: Outputs are always written asynchronously

Gain Drift: 10 ppm/˚C max
Offset Drift
2.5 V Excitation: 0.6 µV/V per ˚C
3.3 V Excitation: 0.5 µV/V per ˚C
5 V Excitation: 0.3 µV/V per ˚C
10 V Excitation: 0.2 µV/V per ˚CChannel-to-Channel Matching (Calibrated)
Input Signal
Frequency (fin)TypicalGain
Maximum

0 to 1 kHz

0.15%

0.3%

1 to 20 kHz

0.4%

1.1%

Phase Nonlinearity
0 to 1 kHz: <0.001˚
0 to 20 kHz: ±0.1˚
Input Delay: 4.8 µs + 38.4/fs
PassbandFrequency: 0.45 • fs
Flatness: 0.1 dB max
StopbandFrequency: 0.55 • fs
Attenuation: 100 dB
Alias-Free Bandwidth: 0.45 • fs
Oversample Rate: 64 • fs
Attenuation at Oversample Rate1
		 50 kS/s: 90 dB @ 3.2 MHz
		 10 kS/s: 60 dB @ 640 kHz
Common-Mode Voltage
All Signals to Earth Ground: ±60 VDC
Common-Mode Rejection Ratio (CMRR)
Relative to Earth Ground2: -140 dB @ 0 to 60 Hz
Relative to EX–: -85 dB @ 0 to 1 kHz

1
2

Rejection by analog prefilter of signal frequencies at oversample rate.
Measured with a balanced cable. Shielded cables may be significantly unbalanced.

Measurement Computing(508) 946-5100

4

info@mccdaq.com

mccdaq.com

6224

Specifications and Ordering Information
Ordering Information

Counters
Channels: Up to 4 independent
Resolution: 32-bit
Input Frequency: 20 MHz max
Input Characteristics: 10 kOhm pulldown
Trigger Level: TTL
Minimum Pulse Width: 25 ns high, 25 ns low
Programmable Modes: Counter, encoder
Encoder Resolution: x1 (default), x2, and x4
Encoder Sources: There are 3 encoder sources (A, B, and Z) that be assigned toany digital pin x
Counter Source: Internal clock, timer 1, timer 2, and digital pin x; one source canbe used in multiple counters
Counter Mode Options: Totalize, clear on read, rollover, stop at the top, increment,
decrement, rising edge, falling edge
Counter Gate Options: Unused, internal clock, timer 1, timer 2, and digital pin x;
one gate can be used in multiple counters

Description
12-channel, 24-bit resolution, Ethernet-based strain measurement
module, with included Out-of-the-Box Encore software

Part No.
6224

Accessories & Cables
Universal power supply, 24 VDC @ 0.8 A (max); requires additional cable,
CA-1 (US) or CA-216 (European)
TR-60U
Required cable for use with TR-60U; 120 V US version
CA-1
Required cable for use with TR-60U; 220 V European version
CA-216
1
RJ12 shielded cable, 6 conductor, SYNC, 0.3 m (1 ft.)
CA-74-1
RJ50, 12-pin screw-terminal connector, 4 packCN-268
RJ50, 120 Ohm quarter bridge connector, 4 packCN-269
RJ50, 350 Ohm quarter bridge connector, 4 packCN-270
RJ50 to RJ50 M/M, 1 m (3.28 ft.) cable, 4 packCA-272-01
CA-192-7C
Ethenet crossover cables, 2.133 m (7 ft.) 2, 3
Ethernet patch cable, 0.457 m (1.5 ft.)2
CA-242Ethernet patch cable, 2.133 m (7 ft.)2
CA-242-7
Stacking plate kit
190658A-01
Handle kit
HA-210-5-BK

Power
Power Consumption: 6.3 W typ, 6.6 W max
Power Jack: Barrel type, 5.5 mm O.D.; 2.1 mm I.D.

Physical Characteristics
Weight: 1.3 kg (2.88 lbs)
Dimensions: 276.9 mm W x 169.8 mm D x 30.5 mm H (10.9” x 6.685” x 1.2”)

Environmental
6224 units are intended for indoor use only but may be used outdoors if installed
in a suitable enclosure.
Operating Temperature: -40 to 50 ˚C
Storage Temperature: -40 to 75 ˚C
Ingress Protection: IP 40
Operating Humidity: 10 to 90% RH, noncondensing
Storage Humidity: 5 to 95% RH, noncondensing
Maximum Altitude: 2,000 m (6,562 ft.)
Pollution Degree: 2

Calibration
Calibration Interval: 1 year; contact factory for information regarding calibration service.
Note: The above calibration information pertains to hardware calibration, notto be confused with “user” or “software” calibration. When performing a “user”
calibration via Encore (or other software) keep in mind that sample rate affects
both gain and offset of the hardware, and therefore any “user” calibration
should be performed at the same sample rate that is intended for measurements.

1 Up to nine units can be synchronized. The total combined length of the SYNC
cables is not to exceed 2.438 m (8 ft.).
2 Ethernet cable length must be <3 m (9.8 ft.) in order for the system to be CE
Compliant.
3 Ethernet crossover cables should only be used for direct network connections.
In particular, attempting to connect a device to a Hub using a crossover cable
may prevent that network link from functioning. Some modern routers havebecome an exception by including logic to detect the crossover cable and allow
the network link to function.

Measurement Computing(508) 946-5100

5

info@mccdaq.com

mccdaq.com

6230 & 6231

High-Speed, Isolated Voltage Input Modules
Features• 12 voltage inputs• 24-bit ADC per channel, 50 kHz sample
rate• ±10 V (6230) or ±60 V (6231) input range• 250 V channel to channel isolation• Simultaneous sampling• Eight digital I/O• Screw-terminal connectors• Multiple trigger modes• Ethernet-interface to PC• Expandable using multiple 6000 Series
modules

The 6230 and 6231 with Encore software are high-speed isolated voltage measurement solutions

Software• Includes Encore interactive measurement software for Out-of-the-Box
setup, acquisition, display, logging,
analysis, and reporting• Advanced feature set with no programming required• Drag-and-drop functionality• Easy-to-use set-up wizards• Ability to drag data directly to Excel®
• Included data analysis tools• Playback mode
• Sophisticated test report capability• Incorporate multiple 6000 Series
devices into one application• Supported Windows® Operating
Systems: Windows 7 (32-bit or 64-bit),Windows Vista (32-bit or 64-bit),Windows XP SP2 (32-bit),Windows 2000 SP4

Overview
The 6230 and 6231 are high-speed, isolated
voltage input modules. The 6230 includes 12
voltage inputs with a ±10 V range. The 6231
offers a ±60 V range. Each module features
a 24-bit ADC per channel and channel-tochannel isolation. The 6230 and 6231 arepart of the 6000 Series of Ethernet-based
data acquisition modules which feature
integrated signal conditioning and a modular
design that allows for system expansion.
Because they use an Ethernet connection,
modules can be connected directly to a PC
or used in remote configurations utilizing
multiple modules. Each 6000 Series module
includes Encore interactive measurement
software. Encore couples ease-of-use withadvanced functionality including custom
data displays, powerful analysis, and detailed
reporting capabilities.
Measurement Computing

6000 Series Module Overview
6220

6222

6224

6230

6231

FeatureAnalog input typevoltagethermocouple

strain gage/bridge

voltagevoltage

Analog inputs

12

12

12

12

12

Digital I/O

8

8

8

8

8

Resolution

16 bit24 bit24 bit24 bit24 bitSample rate

50 kHz per channel

100 kHz per channel

2 Hz

50 kHz per channel

50 kHz per channel

Voltage input range±10 V±80 mV±25 mV/V

±10 V

±60 V

Connector type

BNC

screw-terminal

RJ50

screw-terminal

screw-terminal

Ch-to-ch isolation

—

—

—

Yes

Yes

Analog Inputs
The 6230 is equipped with 12 analog input
channels with a ±10 V range. The 6231
includes a ±60 V range. Each analog input
features a 24-bit ADC with a maximum
50 kHz sample rate per channel. All
channels are sampled simultaneously.
The 6230 and 6231 also include 250 Vrms
channel-to-channel isolation for increased
signal integrity and protection.

Digital I/O
The 6230 and 6231 include eight digital
I/O lines. All eight lines are accessible via
a 9-pin female DSUB connector located on
the rear panel of the unit. Each digital I/O
bit can be programmed individually to beeither an input or an output.

Encore provides real-time viewing andanalysis of data along with built-in test
report capabilities

Counters

Triggering

There are four 32-bit counters built into
each 6000 Series module. They are accessed
through the 9-pin female DSUB connector located on the rear of the unit. Eachcounter has a maximum 20 MHz input(508) 946-5100

1frequency and can be used in counter or
encoder (A, B, and Z) modes.

A variety of trigger modes are supportedby the 6230 and 6231. A wide selectionof programmable analog and digitaltrigger modes are available for startingan acquisition. All trigger modes, along

info@mccdaq.com

mccdaq.com

6230 & 6231
General Information

6230/6231 System Examples
Notebook PC running
Encore software

Notebook PC running
Encore software

Ethernet

Ethernet network

6230 high-speed
voltage input module

Two 6230 high-speed voltage
input modules

with the number of scans and the sample
rate for pre- and post-trigger data, aresoftware programmable prior to the startof a scan sequence.

6230 and 6231 Block Diagram
To Ch 1-4
To Ch 5-8

System Power Connection

CH1 TC

Ethernet Features
The 6000 Series transfers acquired data
to the PC via 10/100BaseT Ethernet,
allowing a continuous stream of data to
be collected and stored in a PC’s memoryor hard drive. The most common andhighest-performance connection is withdedicated, point-to-point Ethernet link
between the PC and the 6230 and 6231.
With an enterprise-wide Ethernet network
connection, any number of 6000 Series
modules can be connected to the network.

CH2 TC

CH3 TCCH4 TC–+–+Internal Circuit(Channels 1-4) – See Note
Diff
Amp

24-BitADC

Isolator

Removable
Screw-Terminal
Connector

CH1

Diff
Amp

24-BitADC

CH2 Control
Logic(Ch 1-4)–

CH3 Circuit*

+
–

CPU

10/100 MB
Ethernet

PHY
Data
Buffer
MemoryCH3CH4CH4 Circuit*+

Acquisition Controller

To Ch 9-12

Isolator

The 6230 and 6231 offer the flexibility to
be powered either directly from a 19 V to
30 VDC source, or via the included TR-60U
AC power adapter.

Power
SupplyCalibration
Memory* Channel 3 and 4 circuitry (not illustrated) looks like the circuitry
for channels 1 and 2.
LEDs
Internal Circuit (Channels 5-8) – See Note

Multi-Unit Synchronization

Measurement Computing(508) 946-5100

Digital I/O

SYNC
Ports
Internal Circuit (Channels 9-12) – See Note

Acquisition Controller

Multiple 6000 Series modules can besynchronized via the rear-panel SYNC
ports on each unit. After connecting
each module to an Ethernet port, simply
connect multiple 6000 Series modules
together using SYNC cables (CA-74-1).
Encore software provides the capability ofsetting one of the 6000 Series modules as
the master and the others as slaves. The
sampled data phase relationship amongchannels between multiple devices isdependent on the “Channel SYNC Skew”
specification for each device.

Chassis#6-32

Note: The schematics for channel group 5 through 8 and group
9 through 12 look like that of the first group. Each channel has its own
differential amplifier and ADC. All channels are electrically isolated.

2

info@mccdaq.com

mccdaq.com

6230 & 6231
Specifications
Specifications

The following specifications are typical for the temperature range -40 to 50 ˚Cunless otherwise noted. The specifications apply to both the 6230 and 6231 except
when otherwise noted.

Input Characteristics
Number of Analog Channels: 12
ADC Resolution: 24 bits
Type of ADC: Delta-Sigma (with analog pre-filtering)
Sampling Mode: Simultaneous
Data Rate Range (fs)Minimum: 1.613 kS/s
Maximum: 50 kS/s
Data Rates (fs): (50 kS/s)/n , n = 1, 2, …31
Multiple Device, Channel Sync Skew1: sample period
Single Device, Channel-to-Channel Matching (Calibrated)
6230: 208 nS (max)6231: 125 nS (max)Master Timebase (Internal)
Frequency: 12.8 MHz
Accuracy: ±100 ppm max

Alias-free bandwidth: 0.453 • fs
-3 dB prefilter bandwidth (fs  = 50 kS/s): 24.56 kHz
CMRR (fin  = 60 Hz)
6230: 126 dB
6231: 116 dB
SFDR (1 kHz, –60 dBFS): -128 dBFS
Total Harmonic Distortion (THD)
1 kHz, -1 dBFS: -99 dB
1 kHz, -20 dBFS: 105 dB

Operating Voltage Ranges
Measurement Voltage, AI+ to AIDevice

Nominal (V)Typical (V)Minimum (V)

6230

±10

±10.52

±10.3

6231

±60

±62.64±61.5

Overvoltage Protection: ±100 V
Input Coupling: DC
Input Impedance (AI+ to AI-): 1 MOhm

Digital I/O
Channels: 8 Digital I/O, programmable as a single port, or as individual lines
Power-Up Mode: Inputs pulled low
Connector: DB-9 female		
Programmable Input Scanning Modes
Asynchronous: Under program control at any time relative to analog scanning.
Synchronous: Data captured synchronously with the analog channels.
Input Levels
Low: 0 to +0.8 V
High: +2.0 V to +5.0 V
Input Voltage Range without Damage: -0.6 V to +5.6 V max
Input Pull-Down Resistor: 10 kOhm
Output Voltage Range: 0 to +3.0 V(may be externally pulled up to 5.6 V without damage)
Output Resistance: 40 Ohm
Output Levels: (see 6000 Series Digital Output graph above)
Sampling: 1 MHz max continuous
Output Timing: Outputs are always written asynchronously.

Accuracy, 6230 (Typical Range: ±10.52V)
Measurement Conditions

Percent of Reading(Gain Error)

Offset Error

Calibrated max (-40 to 50 ˚C)

0.13%±6.31 mV

Calibrated typ (25 ˚C, ±5 ˚C)

0.03%±842 µV

Accuracy, 6231 (Typical range: ±62.64V)
Measurement Conditions

Percent of Reading(Gain Error)

Offset Error

Calibrated max (-40 to 50 ˚C)

0.13%±31.3 mV

Calibrated typ (25 ˚C, ±5 ˚C)

0.03%±5.01 mVCounters

Input Noise6230: 70 µVrms
6231: 320 µVrms
Stability
Gain Drift: ±5 ppm/˚C
Offset Drift		 6230: ±24 µV/˚C
		 6231: ±150 µV/˚C
Post Calibration Gain Match (Ch-to-Ch, 20 kHz): 0.22 dB max
Crosstalk (1 kHz): -130 dB
Phase Nonlinearity (fS = 50 kS/s): 0.11˚ max
Input Delay
6230: 38.4/fS + 3 µs
6231: 38.4/fS + 2.6 µs
PassbandFrequency: 0.453 · fS
Flatness (fS = 50 kS/s): ±100 mdB max
StopbandFrequency: 0.547 • fs
Rejection: 100 dB

Measurement Computing(508) 946-5100

Channels: Up to 4 independent
Resolution: 32-bit
Input Frequency: 20 MHz max
Input Characteristics: 10 kOhm pulldown
Trigger Level: TTL
Minimum Pulse Width: 25 ns high, 25 ns low
Programmable Modes: Counter, encoder
Encoder Resolution: x1 (default), x2, and x4
Encoder Sources: There are 3 encoder sources (A, B, and Z) that be assigned toany digital pin x
Counter Source: Internal clock, timer 1, timer 2, and digital pin x; one source canbe used in multiple counters
Counter Mode Options: Totalize, clear on read, rollover, stop at the top, increment,
decrement, rising edge, falling edge
Counter Gate Options: Unused, internal clock, timer 1, timer 2, and digital pin
x; one gate can be used in multiple counters

1 Mclk is the master timebase.

3

info@mccdaq.com

mccdaq.com

6230 & 6231
Ordering Information

Ordering Information

Power
Power Consumption: 5.5 W typ, 6 W max
Power Jack: Barrel type, 5.5 mm O.D.; 2.1 mm I.D.

Description
Part No.
12-channel, 24-bit ADC, Ethernet-based high-speed ±10 V isolated
voltage input module, with included Out-of-the-Box Encore software6230
12-channel, 24-bit ADC, Ethernet-based high-speed ±60 V isolated
voltage input module, with included Out-of-the-Box Encore software
6231

Safety Voltages (Isolation)
Channel-to-Earth Ground (Continuous): 250 Vrms, Measurement Category II
Channel-to-Channel (Continuous): 250 Vrms, Measurement Category II

Accessories & Cables

Physical Characteristics

Universal power supply, 24 VDC @ 0.8 A (max); requires additional cable,
CA-1 (US) or CA-216 (European)
TR-60U
Required cable for use with TR-60U; 120 V US version
CA-1
Required cable for use with TR-60U; 220 V European version
CA-216
RJ12 shielded cable, 6 conductor, SYNC, 0.3 m (1 ft.)1
CA-74-1
CA-192-7C
Ethenet crossover cables, 2.133 m (7 ft.) 2, 3
Ethernet patch cable, 0.457 m (1.5 ft.)2
CA-242Ethernet patch cable, 2.133 m (7 ft.)2
CA-242-7
Stacking plate kit
190658A-01
HA-210-5-BK
Handle kit

Weight: 1.24 Kg (2.7 lb)
Dimensions: 276.9mm W x 169.8mm D x 30.5mm H (10.9” W x 6.685” D x 1.2”H)
Screw-Terminal Wiring: 16 to 28 AWG copper conductor wire with 7 mm (0.28 in.)of insulation stripped from the end
Torque for Screw Terminals: 0.22 to 0.25 N • m (1.95 to 2.21 lb • in.)Ferrules: 0.25 mm2 to 0.5 mm2 (optional, not supplied)

Environmental
The 6230 and 6231are intended for indoor use, but may be used outdoors if installed
in a suitable enclosure.
Operating Temperature: -40 to 50 ˚C
Storage Temperature: -40 to 75 ˚C
Ingress Protection: IP 40
Operating Humidity: 10 to 90% RH, noncondensing
Storage Humidity: 5 to 95% RH, noncondensing
Maximum Altitude: 2,000 m (6562 ft.)
Pollution Degree: 2

Calibration
Calibration Interval: 1 year; contact factory for information regarding calibration service.
Note: The above calibration information pertains to hardware calibration, notto be confused with “user” or “software” calibration. When performing a “user”
calibration via Encore (or other software) keep in mind that sample rate affects
both gain and offset of the hardware, and therefore any “user” calibration
should be performed at the same sample rate that is intended for measurements.

1 Up to nine units can be synchronized. The total combined length of the SYNC
cables is not to exceed 2.438 m (8 ft.).
2 Ethernet cable length must be <3 m (9.8 ft.) in order for the system to be CE
Compliant.
3 Ethernet crossover cables should only be used for direct network connections.
In particular, attempting to connect a device to a Hub using a crossover cable
may prevent that network link from functioning. Some modern routers havebecome an exception by including logic to detect the crossover cable and allow
the network link to function.

Measurement Computing(508) 946-5100

4

info@mccdaq.com

mccdaq.com

EncoreOut-of-the-Box Software
Features• Interactive measurement software for
Out-of-the-Box setup, acquisition, display, logging, analysis, and reporting• Included with 6000 Series Ethernetbased data acquisition modules• Advanced feature set with no programming required• Drag-and-drop functionality• Easy-to-use set-up wizards• Ability to drag data directly to Excel®
• Included data analysis tools• Playback mode
• Sophisticated test report capability• Incorporate multiple 6000 Series
devices into one application• Supported Windows® Operating
Systems: Windows 7 (32-bit or 64-bit),Windows Vista (32-bit or 64-bit),Windows XP SP2 (32-bit),Windows 2000 SP4

Overview
Encore is the premier Out-of-the-Box dataacquisition software package and is included
with each 6000 Series module. Encore combines ease of use with advanced functionality, including customizable data layouts,
powerful analysis, and reporting capabilities.
Instead of having a program to log data,
another to analyze, and a third to develop
report data, Encore includes the functionalityof all three into one package, thus shortening
the learning curve, and saving time and cost.

Configuration
Easy-to-use set-up wizards provide rapid
hardware configuration. Encore allows you
to go from setup to taking measurements
in minutes. Users can select one or moreavailable devices from Encore’s Hardware
Configuration dialog.
Encore allows you to configure the basic
acquisition or generation options, such as
channel selection, acquisition rate, numberof points, and input range. You can also
configure start and stop triggers.
In addition, within the Channel View table,
users can enable (or disable) channels, selectmeasurement type, enable logging, perform
mx+b calculation, and more. Channel View
configurations can also be shared among
multiple projects.
Measurement Computing

Encore offers an intuitive user interface. Data displays can be configured with multiple charts,
overlapping data plots, and user selectable scales.

Measurement
Encore allows you to develop data displays
quickly with an intuitive, drag-and-drop
interface. Encore also offers the ability toeasily change plot colors, overlay channels (including previously recorded data),
and access multiple devices. You can also
display data in a variety of ways including;charts and graphs, meters, gauges, andtanks. Additionally, you can export data
to other formats including Excel, ASCII
and DIAdem.

Triggering
A variety of programmable analog anddigital trigger conditions are available
for starting and stopping an acquisition
within Encore. Analog triggers include
Rising Edge, Falling Edge (with hysteresis), Above Level, Below Level, Inside
Window, and Outside Window. Digital(508) 946-5100

1

Encore can display channel data in several
ways including meters, gauges, tanks andgraphstriggers include Rising Edge, Falling Edge,
Level High, or Level Low. In addition,
the Duration trigger may used to stop an
acquisition after a specified time.

info@mccdaq.com

mccdaq.com

EncoreOut-of-the-Box Software
Analysis

Reporting

Included analysis options make Encore MCC’s most advanced
Out-of-the-Box software package. Features include FFT, power
spectrum, statistics, peak value, and RMS calculation. In addition, you can also create custom formulas using Encore’s Create
Calculated Signal function. Data can also be viewed in the playback work area. Analysis can be computed on live data as well
as logged data (post processing).
Analysis options include:
•
•
•
•
•
•
•
•

Statistics: Standard Deviation, Mean, and Variance
Peak: +Peak, -Peak, and Peak-Peak
RMS
DC
FFT
Power Spectrum
Minimum
Maximum

Export to Excel®

In the Project Report tab you can drag signals from the Signal Explorer,
enter text, change fonts, import images, and print or export reports
to html

Signals can be exported from Encore to Microsoft Excel. When
exporting signals from Encore to Microsoft Excel, the following
information is exported:••••

Encore offers sophisticated, built-in test reporting capability which
provides the ability to present data in a professional manner.
With Encore’s test report capability, users can easily develop
in-depth test reports.

Signal name
Export date and time
Signal units
Data values

Within Encore’s Project Report tab users can drag signals from theapplication directly into the report and then enter text, change
fonts, import images and print or export reports. Once the report
is complete, you even have the ability to acquire new data andeasily update any previous report, without re-creating the report.
The report can be edited throughout the life of the project.

Measurement Computing(508) 946-5100

2

info@mccdaq.com

mccdaq.com

EncoreOut-of-the-Box Software

1

4
5

2
3
6

1

Signal Explorer displays the data sources, computations, snapshots and logs available for your project. It also
allows you to select and drag signals into the Data View. Signals can also be dragged directly Into Excel.

2

Log Window displays a list of a project’s logged data [including snapshots] sorted by the time at which you
recorded the log or took the snapshot.

3

Snapshots allow you to save a record of the current values of any signal in your project. You can use snapshots
as a reference signal to compare data within the same or another project.
Data View tab allows you to view your data by dragging signals directly from the Signal Explorer or the Log Window.

4 The Data View tab can display signals in several formats, including graphs, charts, and various numeric representations. When you drag a signal to the Data View tab, the signal appears in a new or existing display depending onwhether a display showing that type of data already exists.

5 Legend (Graph Legend) lists every signal displayed in the graph and also shows its corresponding plot color.
Channel View is a table-style section which gives you an overview of all your channels and their settings. You can

6 enable (or disable) channels, select the measurement type, enable (or disable) logging, perform mx+b calculation,and more. You can also use the Channel View toolbar to quickly jump to Acquisition Configuration or [system]
Calibration windows.

Measurement Computing(508) 946-5100

3

info@mccdaq.com

mccdaq.com