WBK61 and WBK62

High-Voltage Adapters

Description
The WBK61 and WBK62 are single-channel high-voltage adapters that can be used with WaveBooks, orwith WBK10A expansion modules. In addition, WBK61 and WBK62 can be used with WBK11A,
WBK12A and WBK13A signal conditioning cards.
WBK61 and WBK62, Product Comparison
WBK61

WBK62

Voltage Divider

200:1 fixed

20:1 fixed

Maximum Voltage

1000 Vpeak (on either input reference toearth ground)

100 Vpeak (on either input reference toearth ground)

Maximum DifferentialVoltage

2000 Vpeak (if neither input exceeds
1000 Vp rating to earth ground)

200 Vpeak (if neither input exceeds
100 Vp rating to earth ground)

The channel output connection [for both the WBK61 and WBK62 voltage adapters] is made through a
BNC-to-BNC coupler. Each model has two signal input-connections, one for low-signal input, and the otherfor high-signal input. The input channels are resistively isolated from ground, providing for safe connectionof the test device. Series resistors, for both high and low channels, serve as attenuators and provide amaximum current limit of 100 µA.
The WBK61 and WBK62 include:• Input HI, safety-style banana-jack• Input LO, safety-style banana-jack• 60-inch (152 cm) cables with probe tips and alligator clips• Green stacking banana plug (for analog common)• CN-110 BNC-to-BNC Coupler• CN-111 Banana Plug/Alligator-clip earth-ground cable
The following figure provides a simplified schematic for each model.

WBK61 and WBK62 Schematics
Note 1: Channel input connections are made from the WBK61/62 BNC to a Channel BNC on the WaveBook dataacquisition system or to the WBK10A analog expansion module. A CN-110 BNC-to-BNC Coupler is used tomake the connection. Refer to the Hardware Setup section of this document module for information regarding
channel and ground connections.
Note 2: The variable resistors are factory set at 50 KΩ and are not user-adjustable.

WBK61 and WBK62, High-Voltage Adapters

988296

WBK61 and WBK62, pg. 1

Hardware Setup
Refer to the following steps and figures to connect the high voltage adapter. Note that we will use the term
“WBK61/62” to refer to both WBK61 and WBK62 since the installation of these models is identical.

WARNING
High voltages can cause death or severe injury. Do not connect or disconnect theprobes from WBK61 or WBK62 when the leads are connected to a voltage source.

WARNING
Failure to properly connect a WBK61 or WBK62 to the acquisition device(WaveBook or WBK10A) and to ground will result in unsafe operation.
1.

Connect WBK61/62 to any input channel(s) of the WaveBook or WBK10A module using the supplied
BNC-to-BNC coupler (CN-110). Refer to the following figures.

2.

If connecting only one WBK61/62, connect the green stacking banana plug to analog common (J12)on the WaveBook or WBK10A module.

A WBK61 Being Connected to a Channel 1 BNC

If connecting two or more WBK61/62s, connect the green stacking banana plug of the first WBK61/62 toanalog common (J12) on the WaveBook or WBK10A module. Connect the other WBK61/62 stacking
banana plugs to the adjacent WBK61/62 earth ground connections (see following figure).

WBK61 and WBK62, pg. 2988296

WBK61 and WBK62, High-Voltage Adapters3.

If connecting only one WBK61/62, connect the green banana plug/alligator clip lead (CN-111) fromthe WBK61/62 earth ground connector to the local earth ground.
If connecting two or more WBK61/62s, connect the green banana plug/alligator clip lead (CN-111)
from the last WBK61/62 earth ground connector to the local earth ground. Refer also to step 2 and theabove figure.

4.

Connect the input leads (CA-152) to the WBK61/62 Input HI and Input LO jacks.

5.

Connect the test leads (CA-152) to the circuit under test. You may use alligator clips (CN-109) toconnect test leads.

If desired, set the applicable WaveBook channel(s) to the appropriate scale by setting the mx+b function inthe WaveView program as discussed in the following section, Software Setup.

WBK61 and WBK62, High-Voltage Adapters

988296

WBK61 and WBK62, pg. 3

Software Setup
Depending on your application, you will need to set several software parameters so that WaveView willorganize data to meet your requirements. Of special importance is the correct setting of the scaling factor.
This is done using the mx + b equation.

WaveView Configuration Main Window
with “Customize Engineering Units” dialog boxes for setting Mx +B and Engineering Units**Note: In actual applications only one Customize Engineering Units box is displayed.
To configure units for WBK61 and WBK62, you must use the Customize Engineering Units dialog box toensure the units are correctly scaled. This is done as follows:
1.

Right-click on the Units column (in WaveView’s main window).

2.

Select mx + b in the Units box. The Configure Engineering Units dialog box will appear.

3.

For WBK61: set m to 200 (see previous figure)
For WBK62: set m to 20 (see previous figure)
Reference Notes:➣ For detailed WaveView information, refer to the WaveView Document Module. Thedocument can be accessed directly from the data acquisition CD via the <View PDFs>
button located on the CD’s opening screen.

➣ If you intend to write your own computer programs refer to the Programmer’s Manual(p/n 1008-0901) as needed. Note that the API does not contain functions that are specificto WBK61 or WBK62. The document can be accessed directly from the data acquisition
CD via the <View PDFs> button located on the CD’s opening screen.

WBK61 and WBK62, pg. 4988296

WBK61 and WBK62, High-Voltage Adapters

WBK61 and WBK62 - Specifications
Name/Function:
WBK61, High-Voltage Adapter with Probes, 200:1 Voltage Divider
WBK62, High-Voltage Adapter with Probes, 20:1 Voltage Divider
Number of Channels: 1Dimensions: 83 mm × 61 mm × 19.5 mm (3.25” × 2.375” × 0.77”)
Cables: 60” leads with detachable probe tips and alligator clips
Output Connector: BNC female
Voltage Divider:
WBK61: 200:1 fixed
WBK62: 20:1 fixed
Maximum Voltage
WBK61: 1000 Vpeak (on either input reference to earth ground)
WBK62:

100 Vpeak (on either input reference to earth ground)

Maximum Differential Voltage:
WBK61: 2000 Vpeak (if neither input exceeds 1000 Vp rating to earth ground)
WBK62: 200 Vpeak (if neither input exceeds 100 Vp rating to earth ground)Frequency Characteristics: approximates a single-pole frequency response-3 dB Bandwidth: 200 kHz minimum
WBK61, Effective Voltage Ranges
Range

WaveBook

Bipolar±2000V±1000V±400V±200V±100V*±40V*±20V*

Unipolar

Not Available

WBK62, Effective Voltage Ranges

WBK10ARange

WaveBook

WBK10A±2000V±1000V±400V±200V±100V*±40V*±20V*±10V*

Bipolar

±200V
±100V
±40V
±20V
±10V*±4V*±2V*±200V
±100V
±40V
±20V
±10V*±4V*±2V*±1V*

0 to 2000V
0 to 1000V
0 to 400V
0 to 200V
0 to 100V*
0 to 40V*
0 to 20V*

Unipolar

Not Available

0 to 200V
0 to 100V
0 to 40V
0 to 20V
0 to 10V*
0 to 4V*
0 to 2V**Available through the WBK11A, WBK12A, or WBK13A option cards.

Measurement Errors:
The following values include total system error, i.e., they include errors from WaveBook, WBK10A,
WBK11A, WBK12A, and WBK13A. The value for gain error does not include offset error.
Gain Error:
0.1% FS (unipolar)
0.2% FS (bipolar)
Offset Error:
0.1% FS (unipolar)
0.2% FS (bipolar)

WBK61 and WBK62, High-Voltage Adapters

988296

WBK61 and WBK62, pg. 5

WBK61 and WBK62, pg. 6988296

WBK61 and WBK62, High-Voltage Adapters