Product Features
Product Features• Ultra-low noise floor, purer signal, never miss the small signals• Up to 12 bits resolution for all the models of this series• Max. analog bandwidth of 100 MHz, 4 analog channels, external trigger output (std.) available
for the dual-channel model• Max. real-time sample rate of 1.25 GSa/s• Max. memory depth of 25 Mpts
• Vertical sensitivity range: 500 μV/div to 10 V/div• Max. capture rate of 1,000,000 wfms/s (in UltraAcquire mode)• Digital phosphor display with real-time 256-level intensity grading• Waveform search and navigation function allows you to debug the signal anomalies faster• 7" (1024x600) capacitive multi-touch screen• Brand new Flex Knob brings user-friendly experience• USB Device & Host, LAN, and HDMI interfaces (std.) for all the models of this series• Novel and delicate industrial design, easy to operate• Unique online upgrade
The DHO800 series is RIGOL's new launched high-performance economical digital oscilloscope.
Though compact in design, it has superior performance. It features a capture rate up to 1,000,000
wfms/s (in UltraAcquire Mode), 25 Mpts memory depth, 12 bits resolution, and low noise.
The DHO800 series is a brand new economical digital oscilloscope designed for the vast
mainstream digital oscilloscope market to meet their design, debugging, and test demands.

Page 1RIGOL Probes and Accessories
Supported
Model

Type

Description

Passive High-impedance Probe• Attenuation: 10:1/1:1
Passive Highimpedance Probe
PVP2150• 1X BW: DC to 35 MHz• 10X BW: DC to 150 MHz• Compatibility: All models of RIGOL's digital
oscilloscopes• Attenuation: 10:1/1:1
Passive Highimpedance Probe• 1X BW: DC to 35 MHz• 10X BW: DC to 350 MHz• Compatibility: All models of RIGOL's digital
oscilloscopes

PVP2350• Attenuation: 10:1/1:1
Passive Highimpedance Probe
PVP3150• 1X BW: DC to 20 MHz• 10X BW: DC to 150 MHz• Compatibility: All models of RIGOL's digital
oscilloscopes

High-voltage Single-ended Probe• Attenuation: 1000:1• BW: DC to 40 MHz• DC: 0 to 10 kV DC
High-voltage Probe• AC: pulse ≤20 kVp-p• AC: sine ≤7 kVrms

RP1010H• Compatibility: All models of RIGOL's digital
oscilloscopes• Attenuation: 1000:1• BW: DC to 150 MHz
High-voltage Probe

RP1018H• DC+ACpeak: 18 kV CAT II• ACrms: 12 kV CAT II• Compatibility: All models of RIGOL's digital
oscilloscopes

Page 2ModelType

Description• Attenuation: 100:1• BW: DC to 300 MHz

High-voltage Probe• CAT I 2000 V (DC+AC)• CAT II 1500 V (DC+AC)• Compatibility: All models of RIGOL's digital
oscilloscopes

RP1300H
High-voltage Differential Probe• BW: DC to 70 MHz
High-voltage
Differential Probe
PHA0150• Max. voltage ≤ 1500 Vpp• Compatibility: All models of RIGOL's digital
oscilloscopes• BW: DC to 100 MHz
High-voltage
Differential Probe
PHA1150• Max. voltage ≤ 1500 Vpp• Compatibility: All models of RIGOL's digital
oscilloscopes• 50X BW: DC to 160 MHz

High-voltage
Differential Probe
PHA2150• 500X BW: DC to 200 MHz• Max. voltage ≤ 1500 Vpp• Compatibility: All models of RIGOL's digital
oscilloscopes• BW: DC to 25 MHz

High-voltage
Differential Probe
RP1025D• Max. voltage ≤ 1400 Vpp (DC + AC P-P)
• Compatibility: All models of RIGOL's digital
oscilloscopes• BW: DC to 50 MHz
High-voltage
Differential Probe
RP1050D• Max. voltage ≤ 7000 Vpp (DC + AC P-P)
• Compatibility: All models of RIGOL's digital
oscilloscopes• BW: DC to 100 MHz
High-voltage
Differential Probe
RP1100D

Page 3• Max. voltage ≤ 7000 Vpp (DC + AC P-P)
• Compatibility: All models of RIGOL's digital
oscilloscopes

ModelType

Description

Current Probe• BW: DC to 300 kHz• Maximum Input
AC: ±100 A
Current Probe

AC P-P: 200 A
AC RMS: 70 A

RP1001C• Compatibility: All models of RIGOL's digital
oscilloscopes• BW: DC to 1 MHz• Maximum Input
AC: ±70 A
Current Probe

AC P-P: 140 A
AC RMS: 50 A

RP1002C• Compatibility: All models of RIGOL's digital
oscilloscopes• BW: DC to 50 MHz• Maximum Input
AC P-P: 50 A (non-continuous)
Current Probe

AC RMS: 30 A
• Compatibility: All models of RIGOL's digital
oscilloscopes

RP1003C• Required to order RP1000P power supply.
• BW: DC to 100 MHz• Maximum Input
AC P-P: 50 A (non-continuous)
Current Probe

AC RMS: 30 A
• Compatibility: All models of RIGOL's digital
oscilloscopes

RP1004C• Required to order RP1000P power supply.
• BW: DC to 10 MHz• Maximum Input

Current Probe

RP1005C

AC P-P: 300 A (non-continuous), 500 A(@pulse width ≤ 30 us)
AC RMS: 150 A• Compatibility: All models of RIGOL's digital
oscilloscopes• Required to order RP1000P power supply.

Page 4

ModelType

Description• BW: DC to 2 MHz• Maximum Input
AC P-P: 700 A peaks, non-continuous

Current Probe

AC RMS: 500 A• Compatibility: All models of RIGOL's digital
oscilloscopes

RP1006C• Required to order RP1000P power supply.

4CH Power Supply

RP1000P

Page 5

Power supply for RP1003C, RP1004C, RP1005C,
and RP1006C; supporting 4 channels.

Specifications
All the specifications are guaranteed except the parameters marked with "Typical" and theoscilloscope needs to operate for more than 30 minutes under the specified operation temperature.

Overview of the DHO800 Series Technical Specifications
Overview of the DHO800 Series Technical Specifications
Model

DHO802

Analog Bandwidth(-3 dB)
Rise Time(10% to 90%, typical)

DHO804

DHO812

DHO81470 MHz

100 MHz≤5 ns≤3.5 ns

No. of Analog
Channels

2 + EXT

Sampling Mode

Real-time Sampling4

2 + EXT

4

Two-channel model:
Max. Sample Rate of
Analog Channel

1.25 GSa/s (single-channel[1]), 625 MSa/s (full-channel[3])four-channel model:
1.25 GSa/s (single-channel[1]), 625 MSa/s (dual-channel[2]), 312.5 MSa/s(full-channel[3])Two-channel model:

Max. Memory Depth

25 Mpts (single-channel[1]), 10 Mpts (full-channel[3])four-channel model:
25 Mpts (single-channel[1]), 10 Mpts (dual-channel[2]), 1 Mpts (fullchannel[3])Max. Waveform
Capture Rate

30,000 wfms/s (Vector Mode)

Vertical Resolution

12 bits

Hardware Real-time
Waveform Recording
and Playing

Max. 500,000 frames

Peak Detection

Capture 1.6 ns glitches

LCD Size and Type

7'' capacitive multi-touch screen

Display Resolution

1024x600

1,000,000 wfms/s (UltraAcquire Mode)

Page 6

Vertical System Analog Channel
Vertical System Analog Channel
Input Coupling

DC, AC, or GND

Input Impedance

1 MΩ ± 1%

Input Capacitance

15 pF ± 3 pF

Probe Attenuation
Coefficient

0.001X, 0.002X, 0.005X, 0.01X, 0.02X, 0.05X, 0.1X, 0.2X, 0.5X,
1X, 2X, 5X, 10X, 15X, 20X, 50X, 100X, 150X, 200X, 500X,
1000X, 1500X, 2000X, 5000X, 10000X, 15000X, 20000X, and50000XCAT I 300 Vrms, 400 Vpk (DC + Vpeak)

Maximum Input
Voltage

Remarks

Whether the probe is used or not, the transient overvoltage
is not allowed to occur.
Please use the instrument dedicated for the specified
measurement category (not applicable to CAT II, III, and IV)

Vertical Resolution

12 bits

Vertical Sensitivity
Range[4]500 μV/div to 10 V/div±0.5 V (<500 μV/div)±1 V (≥500 μV/div, ≤65 mV/div)

Offset Range±8 V (>65 mV/div, ≤270 mV/div)±20 V (>270 mV/div, ≤2.75 V/div)±100 V (>2.75 V/div, ≤10 V/div)

Dynamic Range±4 div (12 bits)

Bandwidth Limit(Typical)20 MHz, FULL; selectable for each channel

DC Gain Accuracy[4]DC Offset Accuracy± 1% (>5mV/div, FullScale)± 2% (≤5mV/div, FullScale, Typ.)≤200 mV/div (±0.1 div ± 2 mV ± 1.5% of offset value)>200 mV/div (±0.1 div ± 2 mV ± 1.0% of offset value)

Channel-to-Channel
Isolation≥100:1

ESD Tolerance±8 kV (on input BNCs)

Page 7

Horizontal System--Analog Channel
Horizontal System--Analog Channel
5 ns/div to 500 s/div
Range of Time Base
Fine
Time Base Resolution

100 psTime Base Accuracy±25 ppm ± 5 ppm/year
Pre-trigger-5 div

Post-trigger

1 s or 100 div, whichever is greater

Time Base Delay Range

Delta Time Accuracy±(Time Base Accuracy x Readout) ± (0.001 x Screen Width)± 20 ps

Channel-to-Channel
Skew Correction±100 ns, Accuracy ± 1 ps

Analog Channel-to-Channel Delay(Typical)[5]Horizontal Mode

≤2 ns

YT

Default

XY

Channel 1/2/3/4

SCAN

Time base ≥200 ms/div

ROLL

Time base ≥50 ms/div, available to enter or exit the ROLL
mode by adjusting the horizontal timebase knob

Acquisition System
Acquisition System
Two-channel model:
Max. Sample Rate of
Analog Channel

1.25 GSa/s (single-channel[1]), 625 MSa/s (full-channel[3])four-channel model:
1.25 GSa/s (single-channel[1]), 625 MSa/s (dual-channel[2]), 312.5 MSa/s(full-channel[3])Two-channel model:

Max. Memory Depth
of Analog Channel

25 Mpts (dual-channel[2]), 10 Mpts (full-channel[3])four-channel model:
25 Mpts (single-channel[1]), 10 Mpts (dual-channel[2]), optional: 1 Mpts
(full-channel[3])Page 8

Acquisition System
NormalDefault

Peak
Detection

Capture 1.6 ns glitches

AverageType2, 4, 8, 16…65536 are available for you to choose

Acquisition Mode

UltraAcquire Waveform capture rate up to 1,000,000 wfms/s

Trigger System
Trigger System
Trigger Source

Analog channel (CH1 to CH4), EXT TRIG[6]Trigger Mode

Auto, Normal, Single

Trigger CouplingDC

DC coupling trigger

AC

AC coupling trigger

High
Frequency
Rejection

Cut-off frequency to 120 kHz (internal trigger only)

Low Frequency
Cut-off frequency to 120 kHz (internal trigger only)
Rejection
Noise Rejection

Increases delay for the trigger circuit (internal trigger only),
On/Off

Holdoff Range

8 ns to 10 s

Trigger
BandwidthInternal trigger: analog bandwidth of the oscilloscope

Trigger Sensitivity

Internal trigger: 0.5 div, ≥50 mV/div; 0.7 div (with noise
rejection enabled)
External trigger[6]: 500 mVpp (DC to 100 MHz)

Trigger Level
Range

Page 9Internal trigger: ±4.5 div from the center of the screen
External trigger[6]:±5 V

Trigger Type
Trigger Type
Trigger Type

Edge

Edge trigger, Pulse trigger, Slope trigger, Video trigger, Pattern trigger,
Duration trigger, Timeout trigger, Runt trigger, Window trigger, Delay trigger,
Setup/Hold trigger, Nth Edge trigger, RS232/UART, I2C, SPI
Triggers on the threshold of the specified edge of the input signal. The edge
types can be Rising, Falling, or Either.
Source channel: CH1 to CH4, and EXT[6]Pulse

Triggers on the positive or negative pulse with a specified width. The pulse
width is greater or smaller than a certain value or within a certain time range.
Source channel: CH1 to CH4

Slope

Triggers on the positive or negative slope of the specified time. The slew timeis greater or smaller than a certain value or within a certain time range.
Source channel: CH1 to CH4

Video

Triggers on all lines, specified line, odd field, or even field that conforms to thevideo standards. The supported video standards include NTSC, PAL/SECAM,
480p/60Hz, 576p/50Hz, 720p/60Hz, 720p/50Hz, 720p/30Hz, 720p/25Hz, 720p/
24Hz, 1080p/60Hz, 1080p/50Hz, 1080p/30Hz, 1080p/25Hz, 1080p/24Hz, 1080i/
60Hz, and 1080i/50Hz.
Source channel: CH1 to CH4Pattern

Identifies a trigger condition by searching for a specified pattern. The pattern is
a combination of multiple selected channel sources. The logic pattern of eachchannel is H, L, X, Rising, or Falling.
Source channel: CH1 to CH4

Duration

Triggers when the specified pattern meets the specified duration condition.
The pattern is a combination of multiple selected channel sources. The logic
pattern of each channel is H, L, and X. The duration is greater or smaller than acertain value, or within a certain time range, or outside a certain time range.
Source channel: CH1 to CH4

Timeout

Triggers when duration of a certain event exceeds the specified time. The event
can be specified as Rising, Falling, or Either.
Source channel: CH1 to CH4

Runt

Triggers when the pulses pass through one threshold but fail to pass throughanother threshold.
Source channel: CH1 to CH4

Window

Triggers in a specified window state when the rising edge of the signal crosses
the upper threshold or the falling edge crosses the lower threshold. The
window state can be Enter, Exit, or Time.
Source channel: CH1 to CH4

Page 10

Trigger Type

Delay

Triggers when the time difference between the specified edges of Source A
and Source B meets the preset time. The duration is greater or smaller than acertain value, or within a certain time range, or outside a certain time range.
Source channel: CH1 to CH4

Setup/Hold

When the setup time or hold time between the input clock signal and the data
signal is smaller than the specified time.
Source channel: CH1 to CH4

Nth Edge

Triggers on the Nth edge that appears after the specified idle time. The edge
can be specified as Rising or Falling.
Source channel: CH1 to CH4

RS232/UART

Triggers on the Start, Error, Check Error, or Data frame of the RS232/UART bus(up to 20 Mb/s).
Source channel: CH1 to CH4

I2C

Triggers on the Start, Stop, Restart, MissedACK, Address (7 bits, 8 bits, or 10bits), Data, or Address Data of the I2C bus.
Source channel: CH1 to CH4
Triggers on the specified pattern of the specified data width (4 to 32) of SPI

SPI

bus. CS[7] and Timeout are supported.
Source channel: CH1 to CH4

Search&Navigation
Search&Navigation
Type

Edge, Pulse

SourceAnalog channel

Copy

Copies the search settings from or to the trigger settings mutually, including
threshold setting and search condition settings

Result Display

Displays in event table form; can be exported to the external or internal
memoryTime navigation: navigates to the acquired waveforms in time order.

Navigation

Event navigation: uses the navigation keys to scroll through the event search
results and navigates to the specified event.
Frame navigation: navigates to the specified frame segment in UltraAcquire
mode.

Page 11

Waveform Measurement
Waveform Measurement
Number of
Cursors

2 pairs of XY cursors
Voltage deviation between cursors (∆Y)

Manual Mode

Time deviation between cursors (∆X)
Reciprocal of ∆X (Hz) (1/∆X)

CursorTrack Mode

Fixes Y-axis to track X-axis waveform point's voltage and timevaluesFixes X-axis to track Y-axis waveform point's voltage and timevaluesAuto
Measurement

XY ModeAllows to display cursors during auto measurement
Measures the voltage parameters of the corresponding
channel waveforms in XY time base mode.
X = Channel 1, Y = Channel 2

Number of
41 auto measurements; and up to 10 measurements can be
Measurements displayed at a time.

Auto
Measurement

Measurement
SourceCH1 to CH4, Math1 to Math4

Measurement
Range(Region)Main, Zoom

All
Measurement

Displays 33 measurement items (vertical and horizontal) forthe current measurement channel; the measurement results
are updated continuously.

VerticalVmax, Vmin, Vpp, Vtop, Vbase, Vamp, Vupper, Vmid, Vlower,
Vavg, VRMS, Per. VRMS, Overshoot, Preshoot, Area, and Period
Area.

Horizontal

Period, Frequency, Rise Time, Fall Time, +Width, -Width, +Duty,-Duty, Positive Pulse Count, Negative Pulse Count, Rising Edge
Count, Falling Edge Count, Tvmax, Tvmin, +Slew Rate, and Slew Rate

OthersDelay(A↑-B↑), Delay(A↑-B↓), Delay(A↓-B↑), Delay(A↓-B↓),
Phase(A↑-B↑), Phase(A↑-B↓), Phase(A↓-B↑), and Phase(A↓-B↓)

Page 12

Waveform Calculation
Waveform Calculation
No. of Math Functions

4 math functions available to be displayed at a time

Operation

A+B, A-B, A×B, A/B, FFT, A&&B, A||B, A^B, !A, Intg, Diff, Sqrt,
Lg, Ln, Exp, Abs, AX+B, LowPass, HighPass, BandPass, and
BandStop

Color GradeSupports FFTFFTRecord LengthMax. 1 Mpts (The max. number of the points to be analyzed
for the FFT operation is 1 Mpts.)

Window Type

Rectangular, Blackman-Harris, Hanning (default), Hamming,
Flattop, and Triangle.

Peak Search

A maximum of 15 peaks, determined by the user-defined
threshold and offset threshold

Waveform Analysis
Waveform Analysis

Waveform
RecordingStores the signal under test in segments according to thetrigger events, that is, saves all the sampled waveform data asa segment to the RAM for each trigger event. The maximum
number of the sampled segments reaches 500,000.
SourceAll enabled analog channels

Analysis

Support playing frame by frame or continuous playing;capable of calculating, measuring, and decoding the played
waveforms
Compares the signal under test with the user-defined mask toprovide the test results: the number of successful tests, failed
tests, and the total number of tests. The pass/fail event canenable immediate stop, beeper, and the screenshot.

PassFail
SourcePage 13

Any analog channel

Waveform Analysis
The waveform histogram provides a group of data, showing
the number of times a waveform hits within the defined region
range on the screen. The waveform histogram not only shows
the distribution of hits, but also the ordinary measurement
statistics.

Histogram

SourceAny analog channel, auto measurement item

Type

Horizontal, vertical, and measure

MeasureSampling
Mode

Statistics: Sum, Peaks, Max, Min, Pk_Pk
Histogram: Mean, Median, Mode, Bin width, Sigma, and XScale
Supports all modes, except the Zoom, XY, and ROLL modes
Provides a dimensional view for waveform intensity, color
grade >16, 256-level color scale display

SourceAny analog channel

Color Theme

Temperature and intensity

Sampling
Mode

Supports all modes

Color Grade

Serial Decoding
Serial Decoding
Number of
Decodings

4 protocol types can be decoded and enabled at the same time

Decoding TypeStandard: Parallel, RS232/UART, I2C, SPI

Parallel

Up to 4 bits of Parallel decoding, supporting any analog channel Support userdefined clock and auto clock settings.
Source channel: CH1 to CH4

RS232/UART

Decodes the RS232/UART (up to 20 Mb/s) bus's TX/RX data (5-9 bits), parity(Odd, Even, or None), and stop bits (1-2 bits)
Source channel: CH1 to CH4

I2C

Decodes the address (with or without the R/W bit) of the I2C bus, data, and
ACK.
Source channel: CH1 to CH4

Page 14

Serial Decoding
Decodes the MISO/MOSI data (4-32 bits) of the SPI bus. The available mode
SPI

includes "Timeout" and "CS"[7].
Source channel: CH1 to CH4

Auto
Auto
AutoScale

Min voltage > 10 mVpp, duty cycle > 1%, frequency > 35 Hz

Digital Voltmeter
Digital Voltmeter
SourceAny analog channel

Function

DC, AC+DCrms, ACrms

Resolution

ACV/DCV: 3 digits

Limits Beeper

Sounds an alarm when the voltage value is within or outside of the limit range

High-precision Frequency Counter
High-precision Frequency Counter
SourceAny analog channel and EXT[6]MeasureFrequency, period, totalizer
Resolution

3-6 digits, user-defined

Max. Frequency

Max. analog bandwidth

Counter

48-bit totalizer
Totalizer
Counts the number of the rising edges
Time Reference

Internal reference

Command Set
Command Set
Common Commands Support

IEEE488.2 Standard

Error Message Definition

Error messages

Page 15

Command Set
Support Status Report Mechanism

Status Reporting

Support Syn Mechanism

Synchronization

Display
Display
LCD

7-inch capacitive multi-touch screen, gesture enabled
operation

Resolution

1024×600 (Screen Region) 16:9

Graticule

10 horizontal divisions x 8 vertical divisions

Persistence

Off, Infinite, variable persistence (100 ms to 10 s)

Brightness

256 intensity levels (LCD, HDMI)

Processor System
Processor System
Processor

Cortex-A72 up to 1.8 GHz, 6-core processor

System Memory

4 GB RAM

Operating System

AndroidInternal Non-volatile Memory

8 GB

I/O
I/O
USB2.0 Host

1 on the front panel

USB2.0 Device

1 on the rear panelLAN

1 on the rear panel, 10/100 Base-T, supporting LXI-C

Web Remote
ControlSupports Web Control interface (input the IP address ofthe oscilloscope into the Web browser to display theoperation interface of the oscilloscope)

EXT Interface[6]Trigger Input

1 on the front panel, BNC connector

Page 16

I/O
1 on the rear panel, BNC connector
Output Interface

Vo (H) ≥ 2.5 V open circuit, ≥1.0 V 50 Ω to GND
Vo (L) ≤ 0.7 V to load ≤4 mA, ≤0.25 V 50 Ω to GND

AUX OUTHDMI

Trigger Output

Outputs a pulse signal when the oscilloscope is triggered

Pass/Fail

Output a pulse signal when a pass/fail event occurs.
Supports user-defined pulse polarity and pulse time (100ns to 10 ms)

Video Output

1 on the rear panel, HDMI 1.4, A plug. It is used to connect
to an external monitor or projector

Probe
Compensation
Output1 kHz, 3 Vpp square waveform

Power Supply
Power Supply
Power Supply
Interface

Type-CPower Voltage

DC 12 V, 4 A

PowerMax. 48 W (when connected to various interfaces, USB storage device, active
probes)Environment
Environment
Temperature
Range

Operating

0℃ to 50℃

Non-operating-30℃ to +60℃
below +30℃: ≤90% RH (without condensation)

Operating+30℃ to +40℃, ≤75% RH (without condensation)

Humidity Range+40℃ to +50℃, ≤45% RH (without condensation)
Non-operating

below 60℃: ≤90% RH (without condensation)

Operatingbelow 3,000 m

Non-operating

Below 15,000 m

AltitudePage 17

Warranty and Calibration Interval
Warranty and Calibration Interval
Warranty

Three years for the mainframe, excluding the probes and accessories.

Recommended
Calibration Interval

18 months

Regulations
Regulations
Compliant with EMC DIRECTIVE 2014/30/EU, compliant with or higher than
the standards specified in IEC 61326-1:2013/EN 61326-1:2013 Group 1 ClassACISPR 11/EN 55011

Electromagnetic
Compatibility

IEC 61000-4-2:2008/EN
61000-4-2±4.0 kV (contact discharge), ±8.0 kV (airdischarge)

IEC 61000-4-3:2002/EN
61000-4-3

3 V/m (80 MHz to 1 GHz); 3 V/m (1.4 GHz to
2 GHz); 1 V/m (2.0 GHz to 2.7 GHz)

IEC 61000-4-4:2004/EN
61000-4-4

1 kV power line

IEC 61000-4-5:2001/EN
61000-4-5

0.5 kV (phase-to-neutral voltage); 1 kV
(phase-to-earth voltage); 1 kV (neutral-toearth voltage)

IEC 61000-4-6:2003/EN
61000-4-6

3 V, 0.15-80 MHz

IEC 61000-4-11:2004/EN
61000-4-11

Voltage dip: 0% UT during half cycle; 0% UTduring 1 cycle ; 70% UT during 25 cycles
short interruption: 0% UT during 250 cycles

Page 18

Regulations
EN 61010-1:2019
EN 61010-031:2015
IEC 61010-1:2016
IEC 61010-2-030:2017
SafetyUL 61010-1:2012 R7
UL 61010-2-31:2017 R2
CAN/CSA-22.2 No. 61010-1-12:2017
CAN/CSA-22.2 No. 61010-2-30:2018
CAN/CSA-22.2 No. 61010-031-07:201

Vibration

Meets GB/T 6587; class 2 random
Meets MIL-PRF-28800F and IEC60068-2-6; class 3 random
Meets GB/T 6587-2012; class 2 random

Shock

Meets MIL-PRF-28800F and IEC 60068-2-27; class 3 random
In non-operating conditions: 30 g, half-sine wave, 11 ms duration, 3 shocks
along the main axis, total of 18 shocks

Mechanical Characteristics
Mechanical Characteristics
Dimensions

265.35 mm (W) x 161.75 mm (H) x 77.38 mm (D)Package excluded: 1.78 kg

Weight[8]Package included: 2.78 kg

Non-volatile Memory
Non-volatile Memory
Setup/Image
Data/File Storage

setup (*.stp), image (*.png, *.bmp, *.jpg)

CSV waveform data (*.csv), binary waveform data
Waveform Data (*.bin,), list data (*.csv), and reference waveform data(*.ref, *.csv, *.bin)

Internal Capacity

8 GB

Reference Waveform

Displays 10 internal waveforms

Setting

Storage is limited by the capacity

Page 19

Non-volatile Memory
USB Capacity

Supports the USB storage device that conforms to theindustry standardNote:[1]: Single-channel mode: If any one of the channels is enabled, it is called single-channel mode.
[2]: Dual-channel mode: For four-channel models, if any two of the channels are enabled, it is called dual-channel mode.
[3]: Full-channel mode: For two-channel models, if all of the two channels are enabled, it is called full-channel mode; for fourchannel models, if any three channels or all of the four channels are enabled, it is called full-channel mode.
[4]: 500 μV/div is a magnification of 1 mV/div setting. For vertical accuracy calculations, use full scale of 8 mV.
[5]: For any channel, under the same input impedance with DC-coupled, the Volts/div setting is the same for 100 mV/div and
200 mV/div.
[6]: Only available for the two-channel models.
[7]: Only available for the four-channel models.
[8]: Standard configuration.

Page 20

Order Information and Warranty
PeriodOrder Information
Order Information

Order No.

Model
100 MHz, 1.25 GSa/s, 25 Mpts, 4CH

DHO814100 MHz, 1.25 GSa/s, 25 Mpts, 2CH

DHO81270 MHz, 1.25 GSa/s, 25 Mpts, 4CH

DHO80470 MHz, 1.25 GSa/s, 25 Mpts, 2CH

DHO802Standard Accessories
Power Adaptor Conforming to the Standard of the Destination
Country——

Banana-Plug Ground Connecting Cable——

DHO814/DHO804: Passive Probe x4 (150 MHz)DHO812/DHO802: Passive Probe x2 (150 MHz)

PVP3150

NOTE:
For all the mainframes, accessories and options, please contact the local office of RIGOL.

Warranty Period
Three years for the mainframe, excluding the probes and accessories.

Page 21

Option Ordering and Installation
Process

1. According to the usage requirements, please purchase the specified function options from RIGOLSales Personnel, and provide the serial number of the instrument that needs to install the option.
2. After receiving the option order, the RIGOL factory will mail the paper software productentitlement certificate to the address provided in the order.
3. Log in to RIGOL official website for registration. Use the software key and instruments serial
number provided in the entitlement certificate to obtain the option license code and the option
license file.
4. Download the option license file to the root directory of the USB storage device, and connect theUSB storage device to the instrument properly. After the USB storage device is successfullyrecognized, the Option install menu is activated. Press this menu key to start installing theoption.

Page 22