AN-1403

Rapid Determination of Sr in
50g Soil Samples

Summary of Method Strontium is separated and concentrated from 50 gram soil samples. Soils are leached withconcentrated nitric and hydrochloric acid. The leachate is evaporated to dryness, and the residue is dissolved in 1M
HCl. A ferric hydroxide-calcium phosphate precipitate concentrates strontium and removes matrix components leachedfrom the soil. A calcium fluoride precipitate further concentrates and purifies the strontium fraction. Strontium isseparated from matrix impurities and potentially interfering radionuclides in the sample using two stacked 2mL
cartridges of Eichrom Sr Resin. Radiostrontium is measured on a low background gas flow proportional counter.
Average chemical recovery of strontium, determined by gravimetric yield of stable strontium carrier, is 91 + 4%.
Measured values of 90Sr agreed to within 2% of reference values for 90 minute count times. The minimum detectable
activity for 90Sr in 50g samples with 90 minute count times is 0.41Bq/g. A single operator can prepare batches of 12
samples for the measurement of 90Sr in less than 16 hours.
Figure 1. Sample Preparation
Reagents
Sr Resin, 2mL Cartridges (Eichrom SR-R50-S)
Nitric Acid (70%)
Hydrochloric Acid (37%)
Hydrofluoric Acid (49%) or Sodium Fluoride
Ammonium Hydroxide (listed as 28% NH3 or 56% NH4OH)
Deionized Water
1.25M Ca(NO3)2
3.2M (NH4)2HPO4Strontium Carrier (10mg/mL)
2M Al(NO3)3
Sr-90 standard
Oxalic acid
Boric acid
Equipment

Vacuum Box (Eichrom AR-24-BOX or AR-12-BOX)
Cartridge Reservoir, 20mL (Eichrom AR-200-RV20)
Inner Support Tubes-PE (Eichrom AR-1000-TUBE-PE)
Yellow Outer Tips (Eichrom AR-1000-OT)
50mL and 250mL Centrifuge TubesCentrifuge
Cupped Stainless Steel Planchets (~5mL volume)
Gas Flow Proportional Counter
Muffle Furnace
Hot Plate
Analytical Balance
600mL Glass Beakers
Vacuum Pump

Dry soil at 110°C. Blend and Size.
Remove 50g aliquot into 600mL glass beaker .
Muffle at 550°C for 1-2 hours.
Add 6mg Sr Carrier*, 50mL 70% HNO3,
and 25mL 37% HCl.
*may need to adjust Sr carrier amount toaccount for native Sr content in soil.

Heat to dryness on hot plate, medium setting.
Add 50mL 70% HNO3. Warm sample. Transfer
solids and liquid to 250mL centrifuge tube.
Centrifuge 3500 rpm, 10 min.
Transfer supernate to 600mL beaker.
Add 25mL 70% HNO3 to Solids. Mix andCentrifuge. Transfer supernate to same
600mL beaker. Repeat once.
Discard solids to waste.
Evaporate supernate in 600mL beaker to dryness.
Dissolve residue in 15-20mL 1M HCl.
Transfer to 250mL centrifuge tube.
Dilute to 160mL. Add 1mL 1.25M Ca(NO3)3, 2mL
3.2M (NH4)2HPO4, and 25mL 57% NH4OH. Mix.
Centrifuge. Decant supernate to waste.
Continue to load solution preparation.

Figure 2. Load Solution Preparation and Strontium Separation
Dissolve residue in 40mL 1.5M HCl.
Dilute to 170mL with H2O.
Add 25mL 49% HF.
Mix well. Centrifuge 10 min.
Discard Supernate.
Dissolve residue in 7mL 70% HNO3,7mL 3M HNO3-0.25M Boric Acid,
7mL 2M Al(NO3)3.
(1) Precondition Sr Resinwith 10mL 8M HNO3.
2mLSr(2) Load sample.
Resin(3) Rinse sample tube
with 5mL 8M HNO3.

2mL(4) Add tube rinse to Sr Resin. SrResin(5) Rinse Sr Resinsequentially with:- 15 mL 8M HNO3
- 10mL 3M HNO3 - 0.05 oxalic acid- 10mL 8M HNO3

Gas Flow Proportional Counting*:*(Options for 89/90Sr Discrimination)(8) Evaporate samples to dryness
on tared cupped stainless steel
planchets.
(9) Rinse Sr sample vials with 2mL
0.05M HNO3. Transfer vial rinse toplanchets. Evaporate to dryness.

(a) Sr fraction from step (7) can betransferred to a liquid scintillation vial.
89Sr can be measured by Cerenkov
counting (no LSC cocktail). 89/90Sr canthen be measured by adding liquid
scintillation cocktail.
(b) Sr fraction from step (10) can be
dissolved in 10mL 8M HNO3 after >7days of 90Y ingrowth. 89/90Sr can beremoved on Sr Resin. 90Y will elute in
Sr Resin load and can be counted byliquid scintillation or gas flow
proportional counting.

(10) Weigh planchets on an
analytical balance to determine
gravimetric yield of stable Sr(NO3)2.
(11) Measure radiostrontium insamples on low background gas
flow proportional counter.

(6) Dispose of (1) to (5) as waste.
(7) Strip Sr with 18mL 0.05M HNO3.

Method Performance for 50g Soils Spiked with90Sample
Sr Reference
replicates Value (mBq/g)
75.92
7
11.8
759.2
MDA90

Sr Measured
Value (mBq/g)
5.95 + 0.22
11.5 + 0.7
57.8 + 1.7% Bias
5.0-2.5-2.4

90

Sr
Sr Carrier% Yield
94.0 + 2.6
89.6 + 2.7
89.3 + 4.7

90Sr, 90 minute count, 50g Soil = 0.41 mBq/g

References
1) Sherrod L. Maxwell, Brian K. Culligan, Patrick J. Shaw “Rapid determination of radiostrontium in large soil samples,”J. Radioanal. Nucl. Chem., 295(2), 965-971 (2013).
1955 University Lane Lisle, IL 60532, USA+1 (630)963-0320

www.eichrom.com