DBK162-Channel Strain Gage Expansion CardFeatures• Provides two channels of strain gage
inputs• Accommodates most bridge type sensors, including 4-element full bridges
A 2-channel strain gage expansion cardfor IOtech’s data acquisition systems, theDBK16 is a general-purpose card that accommodates the connection of most strain gage
types, from single-element ­quarter bridges
to 4-element full bridges. The DBK16 also
includes provisions for bridge completion
resistors and pro­­vides four adjustments oneach of its two channels. You can adjust
excitation voltage, input gain, offset nulling, and output scaling. Up to eight DBK16
boards can be connected to one analog
input channel. For applications of five ormore channels, use the 8-channel DBK43A
strain gage module.
Excitation Regulator. The DBK16 accepts
user-supplied excitation input voltage from12 to 16V on each channel. Its on-board
excitation regulator can be adjusted from
1.5 to 10.5 VDC. The regu­­lator’s outputs
have remote sensing terminals and ­feature
50 mA current limiting to prevent ­damage
from short-circuits or overloads. The regulator’s wide voltage range permits you toincorporate virtually any resistive or semiconductive gage type.
Excitation Sources. The 12 to 16 VDC
external excitation source required by theDBK16 can be supplied by the DBK30A
rechargeable battery/excitation module
or by a user-supplied source.
Input Amplifier. The DBK16’s input amplifier provides an input gain range of x100
to x1250. Because input signals can vary
widely in range, the DBK16’s extensive
range provides a better alternative to fixedgain or slot-range amplifiers.
Offset & Scaling Gain. The DBK16’s 0 to
5 VDC offset adjustment range and output
gain scaling permit nulling of large quiescentloads and expansion of dynamic range formaximum resolution. This is an importantfeature because strain gages typically exhibit
pre-load or quiescent output, leading to a
non-zero output prior to the application ofthe load to be measured. The DBK16’s offset
Measurement Computing

See DBK43A for module version of this board

The DBK16 allows connection of most strain gage types to a data acquisition system

DBK16 Bridge Configuration (typical of 2)User suppliedexternalexcitation
power supply(–)

12-16
VDC

1/2 (DBK16)
BypassBypass

Exc.
adj

(+)
8Scaleadj

Excitation
voltageregulator

70.33 µF

LPF
(Av = 2)

Optional
bridgecompletion
resistors

6
Optional Kelvinexcitation connection

5VOUT

A v = A 2 (1-10)+V +5 –V –5

Userstrain
gage
bridgeScale
V2 amp

To LogBook,
DaqBook, or
DaqBoard
analog input
via P1100
Ohm

R23

R24

4

Optional Kelvinexcitation connection+Input
amp–VIN

32
1100Ohm

adjustment is used to null these preload
conditions. The remaining signal can thenbe expanded by the DBK16’s output scaling
amplifier to increase the resolution.

R22

R25Vo

Vi
Av = Ai(100-1250

Offset null
Offset adj.
0-5.0 VDC

Elective capacitive coupling, which is selected via user-installable jumpers, allows
you to separate dynamic signals from static­deflection levels.

Coupling & Filtering. Each channel ofthe DBK16 offers switch-selectable AC or
DC coupling between the input amplifier
circuits and the filter stages. A selectable
3-pole low-pass filter with a user-­cus­tomizable cutoff frequency is also provided.

(508) 946-5100

1

info@mccdaq.com

mccdaq.com

DBK16

Specifications & Ordering Information
Software
The DBK16 includes GageCal™, a
program that enables you to easily
configure, calibrate, and set up yourpressure or load cell application. The
program features an intuitive graphical user interface (GUI) that guides
you through the configuration of
the DBK16 and its adjustment to the­applied strain gage bridge.

You can easily configure the DBK16
via its Windows® software

TheDBK16’ssoftwaresupportsthreecalibration methods: nameplate,2-point, and shunt

Bridge Completion Resistors. Physicallocations are provided for up to four usersupplied bridge-completion resistors perchannel, allowing you to accommodate
virtually any type of external ­configuration
without having to attach bridge ­completion­resistors at the strain gage.

Specifications

Connector: DB37 male, mates with P1*; screw ­terminals
provided for strain gage and external excitation­connections
Number of Channels: 2Excitation Voltage Adjustment Range:
1.50 to 10.50 VDC @ 50 mA
Gain Range: x100 to x1250
Accommodated Bridge Types:
Full Bridge, Kelvin Excitation (6-wire)
Full Bridge (4-wire)
Half Bridge (3-wire)
Quarter Bridge (2-wire)**Bridge Resistors:
On-board locations are provided for four bridgecompletion resistors per channel
Input Type: Differential
Input Impedance: 100 MOhm parallel with 100 pF
CMRR: 115 dB (DC to 60 Hz)
Excitation Voltage Source:
User-supplied 13 to 16 VDC @ 50 mA/channel
Excitation Current Output: 50 mA max
Excitation Sensing: Local or remote
Excitation Regulation
Line Regulation: 0.025%
Load Regulation: 0.05%
Reference Voltages: 2.5 VDC
Reference Accuracy: 0.05%
Reference Drift: 3 ppm/˚CGain Calibration Reference: 5.00 mVDC
Reference Accuracy: 0.2%
Reference Drift: 20 ppm/˚C; separate instrumentation
amplifier for each channel, with gain adjustable fromx100 to 1250 via externally accessible 15-turn trimpot
Gain Accuracy: 0.5%
Gain Drift: 50 ppm/˚C
Input Offset: 100 µV max
Offset Drift: 4 µV/˚C
Output Offset: 20 mV
Offset Drift: 200 µV/˚C
Offset Adjustment: 0 to 100% of range,

0 to 5.00 VDC (15-turn trimpot)
Full-Scale Sensitivity Range
At 5.00 VDC Excitation: 0.8 to 10 mV/V
At 10.00 VDC Excitation: 0.4 to 5 mV/V
Scaling Amplifier Gain Range:
x1 to 10 (15-turn trimpot)
Low Pass Filter: 3 pole, user by-passable-9 dB @ user changeable corner frequency(gain = x2); factory default: 3.5 Hz
Power Consumption:  1040 mW

Ordering Information
Description
2-channel strain gage card

Part No.
DBK16

Accessories
120 Ohm bridge completion resistor
350 Ohm bridge completion resistor
1000 Ohm bridge completion resistor

BCR/120/1
BCR/350/1
BCR/1000/1Cables
For use with DBK10, use CA-37-x ribbon cable, orcontact factory of additional cabling options
For use with DBK60 or LogBook/360, no cable isrequired (except from DBK60 or LogBook/360 tothe A/D mainframe)
For use with no enclosure, use CA-37-x where x is thenumber of DBK devices attached
For use with DaqLab Series (internal slots), use
CA-255-2T with one board, or CA-37-2 for use with
two DBK cards (or contact factory for additionalcabling options)

Product Compatibility✔ LogBook✔ DaqBook✔ DaqLab✔ DaqScan✔ DaqBoard/2000 Series* Attachment to the DaqBoard/2000 Series requires a DBK200, DBK202, DBK203A, DBK209, DBK213, orDBK214** Contact factory for 3-wire connection

Measurement Computing(508) 946-5100

2

info@mccdaq.com

mccdaq.com