MW13/14/16(Training Model, Demonstration Model, Bronchoscopy Model)

Parts change announcement●2019.06

Change on how to attach the replaceable tongue partPreviously, the anchor pin to connect the tip of tongue and the lower jaw was
attached to the tongue side (11392-030). Now the design of the parts has changed
so that the pin is fixed to the jaw side (11392-080). This improvement permits
longer life of the tongue.
Privious model user needs the pin that is fixed to the jaw. Accordinglly, one (1)pin is inclued in this parts set.
Existing part: 11392-030

New part: 11392-080

The pin attached to the lower jaw

The method to assemble and disassemble the new tongue part will differ fromthose found on the instruction manual. Please see the following instructions on
how to attach and remove the tongue part from the lower jaw.
●Instruction on tongue parts replacement(11392-080)

Hook one side of the pin’s head through the hole at the tip of thetongue. Then pull the tongue in the opposite direction to expand
to secure the tongue in place. To remove, pull the tongue to one
side to expand the hole and remove from the pin.
Pull in this direction(when attaching)

tonguePull in this direction(when removing)

Upper view of the pin

holepinpin

Lower jaw

Plastic tube

Lower jaw

Plastic tube

KyotoKagaku America Inc.
USA,Canada,and South America

MW13/14/16(Training Model, Demonstration Model, Bronchoscopy Model)

Change on how to attach the replaceable tongue part2019.08

As of May 2015, the design of the tongue parts and its assembly method has also been changed.
In case you are installing a new type tongue (11392-080) for the first time to the existing system,
the anchor pin need to be fixed to the lower jaw.Follow the instruction below.
＊In case your DAM simulator already has its anchor pin attached to the jaw, the pin do
not need to be replaced.
●Previously, the anchor pin to connect the tip of tongue and the lower jaw was attached tothe tongue side (11392-030). Now the design of the parts has changed so that the pin isfixed to the jaw side (11392-080). This improvement permits longer life of the tongue.
tExisting part: 11392-030

New part: 11392-080

First, ensure that you have received an anchor pin along with the replacement tongue.
Please see the following instructions on how to attach and remove the tongue part fromthe lower jaw.
［Replacement of the tongue parts］
1. Remove the old tongue following the instruction manual.
2. Insert the anchor pin into the holeand push it firmly until the plastic
tube on the leg of the pin touches
the lower jaw. Ensure that the longaxis of the head of the pin is oriented
transversally as shown in the photo.
3. Put the new tongue part into the mouth following the instruction
manual. Then hook one side of the hole on the tongue to the edgeof the pin and pull the tongue in the opposite direction to expand
the hole. Once the hole is large enough, fit the pin through the hole
to secure the tongue in place. To remove, pull the tongue to one
side to expand the hole and remove from the pin.
Pull in this direction(when attaching)

tonguePull in this direction(when removing)

Upper view of the pin

holepinpin

Lower jaw

Plastic tube

Lower jaw

Plastic tube

URL : http://www.kyotokagaku.com
e-mail : rw-kyoto@kyotokagaku.co.jp
KyotoKagaku America Inc.
USA,Canada,and South America

Precautions for use
Thank you for choosing our products.
Though the mask for DAM simulator is a consumption, please pay your attention as follows to prolong the life ofthe mask.
1) Do not push the instruments such as blade of laryngoscope against the corners of mouth because this may
cause the slight tear. The angle of mouth may be torn due to this small tear.

2) As the opening area will be larger, operation will became easier. However, do not push the mask againstmodel teeth and do not pull the mask too strongly. The corners of mouth may be torn.
When assisting to get wide view, pull the corners of mouth with hooking finger on the inner parts. Please referto the following picture.

Inner parts

3) On the training at tight restriction for neck flexibility and mouth opening, do not pull the mask too strongly.
Difficulty of intubation is not changed depending on the setting of the mask.
1. Changing the device 2. Change the procedure4) Apply lubricant to around the mouth and into the mouth. In addition, apply lubricant to the devices you aregoing to use. Operation without applying lubricant occurs friction and damage of the model may be caused.

2018.06