TRC-NW400
Non-Mydriatic Retinal Camera

TRC-NW400

Fully AutomatedTrue Color
Fundus Camera
TRC-NW400

Overview

RoboticFundus Camera

Clear Color Fundus Image,
5MP Image Sensor

Rapid and Simple
Capture by Single touch

Small Footprint. Space Saving

Small Pupil Photography
with φ3.3mm or more

Quick Stereo
Photography

Full 360° rotating monitor
allows operator tooptimize distance from patient

Variety of Connectivity
Options

Connect with your EMR tocreate patient worklists conveniently
in one centralized place

The TRC-NW400 is an automated androbotic fundus camera that performs true color
photography. By incorporating the 360°rotating
touch panel monitor, the operator is able to acquire
the image rapidly and easily by a simple finger
touch. It makes your workflow efficient.

Image Samples
1. Healthy Eye
2. Healthy Eye
3. High Myopia &Glaucoma Suspect
4. Epiretinal Membrane
Image courtesy:
New York Eye and Ear Infirmary of Mount
Sinai NYC

1234

Key Features

High Quality
Image Meets
Ease of Use

Space Saving
The TRC-NW400 incorporates a rotating touch panel monitor that allows the operation of the instrumentfrom virtually any angle. This feature lets the operator be in front, behind or at either side of the patient,optimizing space saving. The TRC-NW400 can comfortably be operated when located against a wall or ina corner.

wallwallOperator

Patient

Fully Automatic Mode

Back Position

In full auto mode, Alignment, Focusing, Shooting and movement between the right eye and left eye areSide Position

Front Position

operated automatically. The operator can use with minimum training. The operator simply touches thescreen to select a mode, touches the center of the pupil and presses “Capture START”.
The TRC-NW400 provides consistent results regardless of operator experience.

Stereo Photography*
In Stereo photography mode, the alignment for a stereo pair is performed automatically. Following theprompts on screen a stereo pair for stereo viewing can be quickly and easily acquired.
* Stereo viewer software is not incorporated.

Register patient

Adjust the chinrest

Result instantly

position and touch

available[Capture START]Manual Operation Mode
In manual mode, the operator can drive the camera using the touch screen as if via the Control lever. Thisis particularly useful in cases when an operator would like to shoot a specific area of the fundus. In manualmode, all movements of the camera are adjusted by finger touch. Illumination level, flash intensity, diopter
adjustment and fixation targets can also be manually regulated.

Register patient

Adjust the chinrest

Stop the auto

Result instantly

position and touchprocess and adjust inavailable[Capture START]“Manual” modeRich Features

Connectivity

Internal Fixation Target Operation

Direct DICOMThe Traditional 3 fixation targets (Disc, Center and Macula) as well as the 9 fixation target for peripheral

The TRC-NW400 is DICOM compliant,photography are incorporated.

making it easy to integrate within PACS

TRC-NW400

DICOM server(Image storage)

and EMR programs.

Workstation
Diagnosis by doctors
LANMWL server

Client PC

Reception PC
Patient info reservation

Shared Folder
DC

The TRC-NW400 can save images intoM

Existing Systema shared folder on a network. By usinga shared folder, the third party’s system
9 Fixation Target

can retrieve the images from the TRC-LANShared folder

TRC-NW400

Other Filing system

NW400 and save them into their system.

Small Pupil Photography& Zoom Function

Ez Capture

The TRC-NW400 enables the user to acquire

The TRC-NW400 includes EZ Capture as

retinal images with pupils Φ3.3mm or more,a standard software.

assisting functions as small pupil aperture, digitalLAN
TRC-NW400

zooming function and overlayed mask.

ExportEz Capture

External Server at Clinic

When the “small pupil mode” is selected, thecamera will automatically adjust for small pupils*.
*Details depend on setting

IMAGEnet 6
The TRC-NW400 can save a digital

Anterior Photography

image to IMAGEnet 6, ophthalmic dataAnterior photography function allows for quicksystem. IMAGEnet 6 is an optional

documentation of external conditions of the eyesoftware.

LANTRC-NW400

IMAGEnet 6 System

TRC-NW400

USB/ HDD

surface and cornea.

Low FlashAnterior Image
Courtesy: New York Eye and Ear Infirmary of Mount Sinai NYC

Direct Storage

For patient comfort and compliance, a low 4.0Ws

Direct Storage is the simplest

flash intensity is normally utilized, producing

configuration and does not require an

excellent images with low flash and reducing

external PC. Convenient for mobile

patient’s discomfort.

stations or when a network is not used.

Reading Center

Specification of TRC-NW400
Observation & photography of the fundus
Angle of coverage

45°/30° (digital zoom)

Working distance

34.8mm (in fundus photography)
φ 4.0mm or more
Small pupil diameter: φ 3.3mm or moreInternal fixation target:
・Dot matrix type organic ELD display .
・The display position can be changed and adjusted.
・The displaying method can be changed.
Peripheral fixation target:
・This is displayed according to the internal fixation
target displayed position.
External fixation target

Pupil diameter for photography

Fixation target

Patient Diopter Correction Range
Without the diopter compensation lens

-13D to +12D

When the concave compensation lens is used

-33D to -12D

When the convex compensation lens is used

+11D to +40D

Power Source
Source Voltage

AC100-240V

Power Input

120VA

Frequency

50-60Hz

Dimensions and Weight
Dimensions

272 - 383mm (W) ×464 - 660mm (D) × 512 - 700mm (H)Weight18kg

Tablet Control

Optional Accessory

External fixation target
The breath shield* is not part of the productconfiguration of TRC-NW400

EF-2*Please check with your local distributor for availability inyour country.

TOPCON CORPORATION
75-1 Hasunuma-cho, Itabashi-ku, Tokyo 174-8580, JAPAN. Phone: +81-(0)3-3558-2522/2502 Fax: +81-(0)3-3965-6898 www.topcon.co.jp
TOPCON MEDICAL SYSTEMS, INC.
111 Bauer Drive, Oakland, NJ 07436,
U.S.A.
Phone: +1-201-599-5100
Fax: +1-201-599-5250
www.topconhealthcare.com
TOPCON CANADA INC.
110 Provencher Avenue, Boisbriand, QC
J7G 1N1 CANADA
Phone: +1-450-430-7771
Fax: +1-450-430-6457
www.topcon.ca
TOPCON HEALTHCARE SOLUTIONS, INC.
111 Bauer Drive, Oakland, NJ 07436,
U.S.A.
Phone: 1-201-599-5100
Fax: 1-201-599-5250
www.topconhealthcare.com
TOPCON HEALTHCARE SOLUTIONSASIA PACIFIC PTE. LTD.
1 Jalan Kilang Timor, #09-01
Pacific Tech Centre, SINGAPORE 159303
Phone: +65-68720606
E-mail: medical_sales@topcon.com.sg
www.topcon.com.sg
TOPCON HEALTHCARE SOLUTIONSEMEA OY
HQ & Product Development
Saaristonkatu 2390100 Oulu, Finland
Phone: +358 20 734 8190
www.topconhealth.eu

TOPCON HEALTHCARE SOLUTIONSAUSTRALIA PTY LTD
14 Park Way, Mawson Lakes,South Australia, 5095, Australia
Phone: +61-8-8203-3306
Email: au.info@topcon.com
www.topconhealth.com.au
REP
TOPCON EUROPE MEDICAL B.V.
Essebaan 11, 2908 LJ Capelle a/d IJssel,P.O.Box 145, 2900 AC Capelle a/d IJssel,
THE NETHERLANDS
Phone: +31 -(0)10-4585077
Fax: +31 -(0)10-4585045
E-mail: medical@topcon.com
www.topconhealthcare.eu

EC

TOPCON ITALY
Viale dell’ Industria 60, 20037 Paderno
Dugnano, (Milano), ITALY
Phone: +39-02-9186671
Fax: +39-02-91081091
E-mail: info@topcon.it
www.topcon-medical.itTOPCON DANMARK
Praestemarksvej 25, 4000 Roskilde,
DANMARK
Phone: +45-46-327500
Fax: +45-46-327555
E-mail: info@topcon.dk
www.topcon-medical.dk

TOPCON IRELAND MEDICAL
Unit 292, Block G, Blanchardstown,
Corporate Park 2 Ballycoolin
Dublin 15, D15 DX58, IRELAND
Phone: +353-12233280
E-mail: medical.ie@topcon.com
www.topcon-medical.ie

TOPCON (GREAT BRITAIN) MEDICAL LTD.
Topcon House, Kennet Side, Bone Lane,
Newbury, Berkshire, RG14 5PX, UK
Phone: +44-1635-551120
Fax: +44-1635-551170
E-mail: medical@topcon.co.uk
www.topcon-medical.co.uk

TOPCON DEUTSCHLAND MEDICAL G.M.B.H.
Hanns-Martin-Schleyer Strasse 41, D-47877
Willich, GERMANY
Phone: +49-(0)2154-8850
Fax: +49-(0)2154-885177
E-mail: info@topcon-medical.de
www.topcon-medical.de

TOPCON FRANCE MEDICAL S.A.S.
1 rue des Vergers, Parc Swen,
Bâtiment 2, 69760 Limonest, FRANCE
Phone: +33-4-37581940
Fax: +33-4-72238660
Email: topconfrance@topcon.com
www.topcon-medical.fr

TOPCON SCANDINAVIA A.B.
Neongatan 2, P.O.Box 25; 43151 Mölndal,SWEDEN
Phone: +46-(0)31-7109200
Fax: +46-(0)31-7109249
E-mail: medical@topcon.se
www.topcon-medical.se

TOPCON POLSKA SP. Z. O. O.
ul. Warszawska 23,
42-470 Siewierz, POLAND
Phone: +48-(0)32-6705045
Fax: +48-(0)32-6713405
Email: info@topcon-polska.pl
www.topcon-medical.pl

TOPCON ESPAÑA S.A.
Frederic Mompou, 4, 08960 Sant Just
Desvern Barcelona, SPAIN
Phone: +34-93-4734057
Fax: +34-93-4733932
E-mail: medica@topcon.es
www.topcon-medical.es

TOPCON SINGAPORE MEDICAL PTE. LTD.
100G Pasir Panjang Road, #05-05,
Interlocal Centre, SINGAPORE 118523
Phone:+65-68720606
Fax:+65-67736150
E-mail: med.sales.sg@topcon.com
www.topcon.com.sg

IMPORTANT In order to obtain the best results with this instrument, please be sure to review all user instructions prior to operation.
Not available for sale in all countries. Please check with your local distributor for availability in your country.
CAUTION : Federal law restricts this device to sale by or on the order of a physician.
©Topcon Corporation | M000039G-4

TOPCON INSTRUMENTS ( MALAYSIA ) SDN.
BHD.
No.6, Jalan Pensyarah U1/28, Hicom
Glenmarie Industrial Park, 40150 Shah Alam,Selangor, MALAYSIA
Phone: +603 5022 3688
Fax: +603 5031 3968
Email : mys_medical_sales@topcon.com
TOPCON INSTRUMENTS ( THAILAND ) CO., LTD.
77/162 Sinnsathorn Tower, 37th Floor,
Krungthonburi Rd., Klongtonsai, Klongsarn,
Bangkok 10600, THAILAND
Phone : 662-4401152
Fax : 662-4401158
Email : tha_medical@topcon.com
MEHRA EYETECH PRIVATE LIMITED
801 B Wing, Lotus Corporate Park, Graham
Firth Steel Compound Goregaon (East)Mumbai 400063 Maharashtra, INDIA
Phone: +91-22-61285455
www.mehraeyetech.in
TOPCON (BEIJING) MEDICAL TECHNOLOGYCO., LTD.
Room 2808, Tower C, JinChangAn Building,No.82, Middle Section of East 4th Ring Road,
Chaoyang District, Beijing 100124, P.R. CHINA
Phone: +86-10-8794-5176