MAX-Serieslnterferometersmaximum usability
maximum accuracy
maximum speed
for Fiber Optic Connector Surface Inspection
Device Features and Benefits

Software Features and Benefits

High resolution optics

One-click operation

Resolve surface details as small as 1.1 μm on single fiber and 1.4 μm on
multi-fiber connectors.

Save time testing large quantities of connectors in Express Mode.

Smart and ergonomic fixtures
Save time with easy ferrule/connector insertion and fixture handling.

Fast measurements
Measure single fiber connector in less than a second and multi-fiberconnector as quickly as 3 seconds with USB 3.0 SuperSpeed connection.

50 μm scan range (white light mode)
Measure large height differences.

High contrast optics
Inspect buffer zones, plastic and metal surfaces.

Wide range of fixtures
Inspect single and multi-fiber connectors, ferrules, bare fibers, MS
termini, both PC and APC.

Modern, portable and robust design
Make high quality measurements in the field.

Microscopy mode
Visually inspect connector endface prior tomeasurement.

MaxInspect™
softwaresupplied with Sumixinterferometers

Customizable administrator and user levelsTune inspection process as per your requirements.

Compliance with industry standards
Test radius of curvature, apex offset, fiber height, core dip and other parameters
per IEC, Telcordia or your own Pass/Fail criteria.

Interactive 2D and 3D surface profiles
Rotate, zoom, scale images to explore surface details.

Configurable report generation and data archival
Document your results your way.

Combined scratch detection and geometry testing
Speed up inspection and increase work efficiency.

Specification
Optimal for connectors
ModelsSingle fiber

BothMulti-fiber

MAX-QS

MAX-QM-B

MAX-QM

Field of view 1.1 × 0.9 mm

3.5 × 1.8 mm

5.3 × 3.0 mm

1.4 µm

2.2 µmBest suitable for ST, FC, SC, MU, LC, E2000, MT-RJ,
connectors/ferrules MS termini(PC and APC)

MT, MTP/MPO, MT-RJ, SMA 905,
MS termini, ST, FC, SC, MU, LC,
E2000

MT, MTP/MPO, MT-RJ, SMA 905

Measurement mode phase shift andwhite lightphase shift andwhite lightwhite light<1 sec*5 sec**

3 sec**

2590 × 1940 px

2590 × 1940 px

external,
via two USB 3.0 cables

external,
via two USB 3.0 cables

3.8 kg (8.4 lbs)

3.8 kg (8.4 lbs)

103 x 137 x 183 mm(4.06 × 5.39 × 7.2 in)

103 x 137 x 183 mm(4.06 × 5.39 × 7.2 in)manual and short range auto

manual and short range auto

Lateral resolution 1.1 µm

Scanning and calculation speed 2 sec*
Maximum image size 2590 × 1940 px
Power supply external,
via two USB 3.0 cables
Weight 3.9 kg (8.6 lbs)
Dimensions 103 x 137 x 183 mm(H × W × L) (4.06 × 5.39 × 7.2 in)Focus manual and short range auto
* 64 MHz, standard single fiber connector** 64 MHz, MT 12 connector with small x, y angles

America
USA EAST 1-888-32-SENKO
USA WEST 1-858-623-3300
TEXAS 1-972-661-9080
Sales-Americas@senko.com
South America
BRAZIL +55-21-3736-7065
Sales-Brazil@senko.com

Asia
HONG KONG +852-2121-0516
SHANGHAI +86-21-5830-4513
SHENZHEN +86-755-2533-4893
Sales-Asia@senko.com

Europe
FRANCE +44 7939364565
Salesfrance@senko.com
GERMANY +49(0)15117683072
Sales-Germany@senko.com
ITALY +39 338 8919089
Sales-Italy@senko.com
POLAND +44 (0) 7796444488
Sales-Europe@senko.com

SPAIN & PORTUGAL +34 678042829
Sales-Iberia@senko.com
UK +44 (0) 1256 700880
Sales-UK@senko.com
OTHER +44(0) 1256 700880
Sales-Europe@senko.com

Asia Pacific
AUSTRALIA +61 (0) 3 9755-7922
Sales-Asia-Pacific@senko.com
Middle East North Africa
DUBAI +971 4 8865160
Sales-MENA@senko.com

Japan
TOKYO +81 (0) 3 5825-0911
Sales-Japan@senko.com