Fast, High-Quality
Imaging

Broadband Probe
Improvement in
Bandwidth

Application Note #134
Survey, Screening, Dynamics:
A No-Compromise Approach to High-Speed
Atomic Force Microscopy
When compared to other common microscopy techniques(optical, SEM, TEM), the atomic force microscope’s (AFM’s)
broad potential for nanoscale imaging and characterization ofnumerous physical surface properties has been somewhat
offset by its slow imaging speed.1 Thus, the AFM has
sometimes been seen as a powerful “specialty tool” to use
when other suitable techniques are not available.
The AFM community has spent considerable effort overthe last decade looking for ways to address the speed
limitation of AFMs, and through this research many of thefundamental technological challenges have been addressedon an academic scale.2-4 Driven by the researcher’squest for discovery, many of these efforts were aimed at
improving the time-resolution of the AFM in order to view
dynamic processes on the nanoscale;2,5 while some also
anticipated the need for the versatility and productivity of afast general-purpose AFM. 3
Bruker’s Dimension FastScan™ development team worked
with many AFM leaders to understand their research
objectives and related enabling technologies, iteratively
exploring various design considerations in the pursuit ofbringing together the best of all these solutions into one
tool. The ultimate result is the creation of an AFM that
ideally marries high-resolution performance with rapid
imaging. This application note details how the Dimension
FastScan AFM accomplishes this ideal.

Sample Dynamics

Applications Requiring Greater AFM Speed
When looking at atomic force microscopy applications thatwould most benefit from improvements in imaging speed,
we found that they could be broadly categorized into threemain areas (see figure 1):
1. The efficient exploration of an unknown,heterogeneous sample, to understand the different
morphologies that best represent the surface, andto finally capture a representative set of images
at publication quality. We call this “survey,” and it
represents the largest of the three areas.
2. The quantitative characterization of a surface property(roughness, number of phases, particle size and shape,
stiffness, etc.) on a large number of samples of thesame class. In this case, the AFM images are only anintermediate; the end product is a graph representing
the measured property versus a parameter of sample
creation (temperature, concentration, stress). We callthis highly applied area “screening.”
3. The observation of sample changes over time, atsufficient speed to time resolve the observed process,where the process can be protein dynamics, aging
phenomena, etc. We call this “classic” area of highspeed atomic force microscopy “dynamics.”

interaction force. In an AFM, there is an inherent tradeoff
between imaging speed and tip-sample force. Any
system can be run incrementally faster when this tradeoff
is utilized. However, it is critical to note that the speed
comes at the expense of increased interaction-force, andis generally limited by either a) the image quality becoming
unacceptable or b) the tip-sample forces becoming
destructive to the tip and/or sample.

Figure 1: AFM Applications that benefit most from higher imaging
speeds break down into the three main categories of survey,
screening, and dynamics.

Though all three areas benefit from increased imaging
speed, they differ in that they each have distinct
requirements. For survey and screening applications, thefocus is on productivity, at high (screening) or sometimes
ultimate (survey) data quality, with no limitation on suchgeneral AFM properties as large and small scan range.
The screening applications also require automation and theloading of multiple samples, while the survey applications
benefit from unconstrained sample format capacity. For
both cases, higher speed must not come at the expense ofincreased operating cost. For dynamics, the priority is frame
rate, at appropriate quality, often with fluid compatibility andgood force control.
To realize the productivity increase of a high-speed AFM,
the overall work flow must also be high speed. If not,another step in the overall data acquisition process (set-up,
sample loading, navigation, engaging, capturing data, andfinal analysis and image processing) immediately becomesthe productivity bottle neck, thus negating the benefits ofhigher tip velocity.

This is still true when building and using a higher speed
AFM. The most general and useful way to describe thespeed increase of an AFM is by the characterization of its
full system transfer function (FSTF) at constant tip-sample
force. Unfortunately, the FSTF is not familiar to many, isdifficult to measure without specialized equipment, and is
hard to use for absolute comparisons due to its dependency
on the force used in the measurement. We therefore findthat a more intuitive way to describe a fast AFM’s speed isby the factor of speed improvement it has over a “normal”
AFM a) on the same samples, b) at the same image quality,
and c) at the same tip-sample force. We call this the“improvement in bandwidth,” as it correlates directly withthe more technical interpretation of this term, when looking
at a property of the AFM’s entire FSTF (see figure 2a).
The AFM tracks the sample surface using a feedback
loop that observes and maintains the interaction of the
AFM probe with the sample surface during scanning byadjusting the tip-sample separation. The components
involved are the AFM probe, photodiode and electronics,
controller, amplifier, and Z actuator. Each component in the
feedback loop introduces its own dynamics (e.g., a delay)
and the sum of all component delays sets an upper limit
for the speed at which the feedback loop can track theaAmplitude
Setpoint–

What is Fast?
Throughout the quest for a faster AFM, a number ofdifferent measures have been used to benchmark the
speed of novel AFM designs: lines/second, frames/second,cantilever resonance frequency, first actuator resonance,
laser spot size, controller sampling rates, etc. This has led tosome confusion when comparing speeds, especially whenusing statements like “video rate” to summarize them.
When considering “normal” AFMs, one finds that theimaging speed at optimal quality strongly depends on thesample, probe, imaging mode, scan size, and interaction
force. (The time needed to complete a frame additionallydepends on the number of lines per image.) The qualityof the AFM image depends on so many factors, because
each of these variables is a contributor to the tip-sample
2

Amplitude
Error+

P/I Controller

Height Data

Integral
Proportional
Gains

Controller
HV Ampifier

Amplitude
Response

Transfer Function: T(s) =Topography
Scanner
Electromechanical
Response

G(s)

–

+

+

Tip/Surface
Distance

SampledAmplitude
SignalProbe
Response

1 + G(s) H(s)

–

G(s)

+

H(s)

b

Figure 2: a) The AFM tracks the sample surface using a feedback
loop. Each component in the loop contributes its own dynamics,e.g., delay. The sum of all component responses, the full speed
system transfer function (FSTF), sets the system‘s limit. b)
Measured FSTF for a typical AFM and of the FastScan System, at
equal tip-sample force. The FSTF rolls off at >20x higher frequency.

BROADBAND-A 27
BROADBAND-B 30
BROADBAND-C 42

Cantilever
The AFM cantilever is, from a simplified physics
perspective, a spring-mass system, with a first resonancefrequency of SQRT (k/m). This is important because (inimaging modes other than contact mode) the cantilever’s
first resonance generally needs to come to equilibriumwith the sample surface in order to provide the informationneeded to track the surface. The settling time this takes
depends on the cantilever’s first resonance frequency(f), divided by its quality factor (Q). Therefore, to makea cantilever for faster imaging, f must be increased or
Q reduced. One can increase f by increasing the spring
constant k, or by reducing the mass m. Increasing thecantilever’s spring constant is not desired, because of ourstated goal to image faster, but at similar tip-sample force.
The mass can be reduced by making the cantilever smaller.
The settling time can be further reduced by reducing Q,
i.e., by increasing the dampening of the cantilever. One
way to do this is to give the cantilever a wide, closed

fr
Q
TSB

ROC

BW ∝

Backside
Coatingk
Q
L (µm)F ∝Tip Height

Before we go into examples in the areas of survey,
screening and dynamics to demonstrate this, we will takea look at the technology improvements that were made ineach AFM system component to achieve the Dimension
FastScan’s overall gain in bandwidth and to utilize it in a
productive work flow.

t (um)

Optimizing AFM System Components

Figure 3 shows Bruker’s Broadband™ A, B and C
probes. These probes are smaller than conventional AFM
probes, and are designed specifically for the operationin the FastScan system. They were developed and are
manufactured by Bruker AFM probes, using Bruker’s
proprietary silicon tip on nitride cantilever process, for thebest combination of flexibility and sharpness.

k (N/m)The other FSTF was measured on the Dimension FastScan
system. The FSTF also rolls off eventually, although
at approximately 20x higher frequency. The expected
improvement in imaging bandwidth is therefore on the orderof 20x.

In the pursuit of high-speed AFMs, smaller cantilevers wererecognized as the key to higher imaging speeds from the
beginning. However, designing a cantilever that balances allthese factors well, while still being able to be manufactured
in large quantities and at reasonable cost, within the tighter
tolerance requirements imposed by smaller cantilevers,
has been one of the biggest obstacles to commercial highspeed AFM technology.

fr (MHz)

Looking at the FSTFs, one can see that the response isflat for low frequencies, and eventually rolls off for higher
frequencies (see figure 2b). What is interesting is that for astandard AFM imaging at 1 line per second and 512 pixels
per line, the pixel frequency would be around 1 kilohertz(with trace and retrace). At this modest rate, the FSTF
shows that the standard system would already have somedifficulty responding to features from pixel to pixel. It is thisconstant pushing of the speed limit that sometimes makes
the force and gain adjustments required to get the best
image quality such an art form.

shape and to reduce tip length, which increases the air
dampening between the cantilever and the sample surface.
Reducing Q would proportionally increase the tip-sample
force per oscillation cycle (because the cantilever comes toequilibrium over fewer cycles and therefore needs to passmore of its kinetic energy to the sample in each cycle). One
therefore needs to further reduce the cantilever’s spring
constant to offset this. This still pays off as Q contributes tothe settling time linearly, while reducing the spring constantk reduces f with a square root dependency.

W (µm)

surface. This sum of the individual components’ behaviors,or transfer functions, is the full system transfer function(FSTF), which determines the AFM’s performance. Speedvariations between different samples are introduced into
the FSTF on the end of cantilever response to the sample,
e.g., the best suited probe and imaging mode, allowabletip-sample force, and smallest sample topography tobe resolved.

32 1.40 18 0.58 2.5 um
32 0.45 1.8 0.3
to 5 nm 5 µm Yes
40 0.21 0.8 0.3 8 um

Figure 3: Scanning faster at similar forces is fundamentally enabled
by smaller cantilevers of equal spring constant. More damping helps
speed but requires further spring constant reduction. Access to
sharp, high-yield probes using economical production processes haslong limited wide adoption of high-speed AFM. Bruker’s Broadband
probes are available in three styles for different application regimes.
SEM micrograph of a Broadband-A probe shown is on the left.

Laser Optics
Operating smaller cantilevers requires an AFM with smaller
laser spot size. The reason all AFMs use a contain verysmall spot size to make them compatible with all sizesof cantilevers is that small spot size AFMs have reduced
optical lever deflection sensitivity. When being used witha shorter cantilever, this effect is cancelled because themechanical lever deflection sensitivity is increased. Thus,
for a given number of nanometer deflection, the angle
change of a shorter cantilever is larger.

3

The reason smaller spot size leads to reduced deflection
sensitivity is optical resolution. To make a smaller spot,
the numerical aperture, or cone angle, of the incoming
laser beam needs to be larger. The cantilever acts as a
mirror, so for a larger, more strongly converging incoming
beam, (to create a smaller spot), the outgoing beam
must be equally divergent. The AFM measures cantileverdeflection by the angle change of the reflected laser beamduring the movement of the laser spot on a four-quadrant
photo detector. The signal generated is proportional to the
movement of this projected spot, divided by the size of the
projected spot. Therefore, a more diverging beam, froma smaller laser spot, results in reduced optical deflection
sensitivity. The consequence is that each cantilever shouldbe operated with an appropriate spot size, i.e., the largest
spot size that will not spill off the edge of the cantilever(which would cause interference and increased noise).
The FastScan head has three different spot sizes thatcan be selected simply by turning a switch. This design
accommodates the broadest range of cantilevers, including
all “normal sized” cantilevers, Broadband probes, and evensmaller (experimental or future commercial) probes, each attheir optimal performance.
Z Scanner and Driver
The second most critical (and challenging) component in
the AFM feedback loop that needs to be tailored for highspeed is the Z scanner. In traditional AFM tube scanners,
the Z scanner was one with the XY scanner. Therefore, it
did not have separate dynamics that could be maximized
independently. Flexure scanners can resolve this issue. In
this case, the (relatively slower) X and Y axis can move a
fast, low-mass, low-inertia Z scanner around. The massdifference of the XY scanner and the mass moved by the Z
scanner can be sufficient to allow the Z scanner to have its
own isolated, fast dynamics. This concept has been used inacademic high-speed efforts. The challenge for a generaluse system is to maintain usability as well as closed-loop
control. From the standpoint of fast dynamics, tip scanningsystems are more robust and less limited than sample
scanning systems, because the mass of the cantileverchip is small and well defined. Sample scanning systems
in general impose restrictions on sample format and mass,and this restriction is multiplied for high-speed systems.
Even for a tip scanning system, the design of a good Z
scanner is challenging. The goal is to maintain a long Z
range, with straight motion, good linearity and integrated
low-noise position sensing, while at the same time allowing
easy loading of cantilevers, and waterproofing of thescanner surfaces for fluid operation and cleaning.
Figure 4 shows the FastScan system’s Z scanner. The
cantilever chip is held by a light-weight, self-contained clip
with sufficient mounting force to ensure good coupling forhigh-frequency (MHz) tapping excitation. The Z motion is
flexure guided. The 3-micron travel is gauged by Bruker’s
proven, ultralow-noise strain gauge technology. The Z
scanner can be removed from the head, and placed on aload stand, for easy cantilever loading.
The exposed surface materials are glass and titanium,
which provide excellent fluid resistance and biocompatibility.
The Z scanner can be thoroughly washed and cleaned for
optimum, low-background DNA and protein imaging in fluid,using a patent-pending wash station design that protects
the Z scanner from the environment while exposing thesurfaces to be cleaned. The wash station doubles as arugged storage/shipping container.
Electronics
The remaining components in the feedback loop areelectronics. On the controller side, the system is based on
Bruker’s NanoScope® V controller, which was designed with

Figure 4: The Z scanner is the second most important component to improve the FSTF for higher speeds. Tip scanning, high-speed designs
don’t restrict sample mass in order to achieve Z scanner performance. The challenge is to increase the Z scanner resonance while maintaining
Z range, linearity/orthogonality, easy probe loading, efficient tapping drive at MHz frequencies, Z sensoring, and fluid compatibility andcleanability. The FastScan Z scanner’s 1st resonance frequency is increased by ~10x, while maintaining all of the above.
4

high speed in mind and performance reserve to support
it. An additional high-performance piezo amplifier drives
the Z piezo at high bandwidth and at high slew rates. The
high slew rate is required to deliver the power to drive highfrequencies also at large amplitudes.
XY Scanner
XY flexure scanners are widely available, and have niceproperties, such as large scan range, flat motion, and goodweight carrying capacity. However, they generally are
not known to be fast. The XY flexure design is desirable
for a fast AFM, as discussed above, to isolate Z scanner
dynamics from the rest of the system. The FastScan uses
an exclusive, patented XY flexure design invented by the
Hansma Lab at UC Santa Barbara, and developed further
by Bruker (see figure 5a).6 The design rigidly couples the
X and Y piezo stacks directly to the scan stage via a set ofparallel flexures from each piezo axis, while maximizing Z
rigidity by directly coupling the scan stage to the frame via
a second set of flexures, to provide flat scanning motion athigh speeds, and suppress any inertial coupling from thefast Z axis.

Figure 5: The FastScan XY scanner (based on a patented design
by Kindt, Fantner & Hansma). The design maximizes out of plane
stiffness to reject coupling of Z-scanner dynamics. One set of blade
flexures (1) connects the stage to the X and Y piezo stacks. Anotherset (2) creates a short, vertically rigid connection to the frame. The
XY scanner combines high Z stiffness, large (>30µm) scan range andflat (<3nm) motion with a first XY resonance frequency >1kHz.

The scanner is capable of 30x30-micron motion (see figure5b), with a 3-nanometer full range flatness (see figure 5c).
The axes are monitored by ultralow-noise strain gauge
position sensors.
Vision Optics
AFMs are typically combined with an optical microscopy
capability to find a sample region of interest, and coarseposition the AFM cantilever above it before engaging. To
realize the gain in productivity promised by faster AFM,
this optical capability must not be compromised and should
provide good optical image quality on a range of opaqueand transparent samples (see figure 6a). The challenge on afast AFM is that the space (and numerical aperture) above
the cantilever is needed both to have a good resolution
optical image, and to project a smaller laser spot. The
solution is therefore to fully integrate the microscope into
the AFM head, and use the objective lens for both imaging,and laser focusing and return.
In doing this, the integrated optical microscope also servesa second purpose: to position and focus the laser spot on
the cantilever (see figure 6b). In the Dimension FastScan
AFM, the optical focus, laser position and four-quadrant
detector offset are all motorized, and are therefore
controllable through the software user interface. The
system automatically focuses onto the cantilever or the
sample surface, depending on the workflow step, prior to
engage, and shows cantilever and sample in focus whenthe AFM is engaged.
Workflow
Much attention was paid to supporting the increased
productivity that results from scanning faster with an

Figure 6: The integrated high-quality digital zoom microscope isused to (a) visualize and navigate the sample (optical images of asemiconductor device and an epithel section, ciliate), and (b) to alignthe laser to the probe (here shown over a 10µm calibration standard).
The software workflow takes care of the switching. Three laser spot
sizes are switch selectable, to support standard, Broadband, andeven smaller probes.

efficient operating workflow. A step-by-step process guides
the user when exchanging the cantilever.
When starting the process, the system moves the stage
into a configuration where it is easy to access and remove
the Z-scanner, which is released from the head by the pushof a button. The flip-over cantilever clip design and side
access grooves make cantilever chip removal and insertion
straight-forward even for occasional users. After inserting
the new cantilever chip, the cantilever type is selected froman extensive built-in database that contains information
about cantilever sensitivity, resonances in different
media, recommended spot size, etc. This information
enables workflow simplifications throughout the software(standardized gains across different cantilevers and spot
sizes, reliable auto tune), and allows the user to work with
SI-units (instead of system-internal arbitrary units) for
many parameters.
The AFM laser and vision optics are con-and par-focal, i.e.,
the AFM laser spot is always in the center of the vision5

optics, and focused on the cantilever when the cantilever
can visually be seen in focus. So, to position the laser spot
on the cantilever, one simply:
1) focuses the cantilever image in the software, using
software buttons,
2) clicks on the cantilever (optionally the laser position
can be fine tuned by clicking “optimize,” or manuallyusing arrow buttons),
3) clicks “zero detector,”4) clicks “auto tune” for TappingMode™.
For imaging in air, this is the entire process to load andadjust the new probe. In fluid, using a “thermal tune”(Fourier spectrum of the thermal cantilever oscillations)is the best method for finding the (very broad) resonancepeak of the cantilever. A new “Fast Thermal Tune” featureoverlays a thermal spectrum over the cantilever sweep in a
few seconds.
Once the system’s imaging bandwidth was increased, it
became apparent that, especially for measurements on
multiple sample sites, the time to engage the probe to thesample became a rate limiting step. In air, the FastScan
uses a new fast engage feature that exploits the effect ofsqueeze-film dampening of the air between cantilever andsample to engage very rapidly (typically in <10 seconds)
to within a few microns of the sample. This new featureis also available for sample navigation, using the cantilever

Figure 7: a) HF etched mica (TappingMode, Air. Probe: Broadband-A). 20µm scan size shows a flat surface with individual etch pits. The
pits correspond to flaws in the lattice that lead to higher local etch rates. b) Detail from inside one etch pit shows the individual mica layers
as “terraces“. The other scan size extreme: Mica lattice (contact mode, Air. Probe: Broadband-B), here imaged at relatively low scan rateand high (1024x1024 pixel) resolution, demonstrating low electrical and mechanical noise, and excellent stability of the automatable, largesample, tip-scanning Dimension Icon/FastScan platform.

Figure 8: Tip life was measured on a rough, hard, highly abrasive TipCheck sample at 17 lines per second. The tip radius was software
estimated from consecutive scans (TappingMode, Broadband-A Probe). The total tip radius increase from 200 TipCheck images is 0.9nm(13%), or 4.5Å/m tip travel.
6

dampening to sense the proximity of the surface and moveit into optical focus. This is particularly useful on very flat,
clean, or transparent sample surfaces (glass, mica), where
good optical references are sometimes hard to find.

situation, and the same level of tip abrasion cannot be
expected for most “normal” samples. However, it does put
a comfortable upper limit on the effect, and illustrates theconsistent low-force tracking performance.

The FastScan system is a tip-scanning design based on
Bruker’s high-end Dimension Icon® platform. Supportedby the optical microscopy features integrated with the
FastScan head, all motorized-stage sample navigation,
focus-to-engage, and automation features of the Icon
platform are available with the FastScan.

Force control is further demonstrated by imaging afragile sample of known morphology, and intricate,steep topography.

Fast Scanning Quality and Force Control
Figure 7a shows a scan of HF-etched mica, taken at 20 lines
per second (i.e., within 12 seconds). The etch pits start witha single flaw in the lattice (induced by natural radioactivity).7
The 20-micron zoom-out demonstrates the FastScan
system’s ability to scan fast over relatively large scan areas(XY scan range = 30 microns), and the flatness of thescan (<3-nanometer bow). The 2-micron zoom-in shows
the individual mica steps taken at 48 Hertz, or 5 seconds/
image (see figure 7b). The data shown is Z sensor data,
illustrating the noise performance of the Z position sensor(0.6 angstroms at 20-Hertz scan rate) on atomic-scale
topography. Figure 7c shows an image of the mica lattice.
This 20-nanometer image was taken at a high resolution of
1024x1024 pixels, and a relatively low scan rate of 6.5 lines
per second. This ultimate-resolution image is testamentthat the fast, large-sample, automatable system capacity
was achieved without the expense of the mechanical noise
and thermal stability expected from a high-end research
AFM platform.

Figure 9 shows an image of a Celgard® polypropylene
battery separator membrane. The membrane is a highlyordered “sieve,” consisting of filaments a few nanometers
in diameter, and larger perpendicular “linkers” (which
also have a sub-structure). The sample is challenging
to image because of the combination of nanometerfilaments supported only at their ends, with deep trenches
in-between. For good judgment of performance, thefilaments should be oriented approximately perpendicular
to the scan direction. To image the trenches the tip needs
to enter tens of nanometers. To come back up onto thefilament the tip-filament interaction must not cause highlateral forces on the filament. The least effect would be a
blurring of the filament edges, and permanent damage tothe filaments can also occur. These effects can be further
judged in detail when looking at the phase image, where
they won’t be masked by the overall topography. Lossof tracking (forces too light to stay in contact) would also
become apparent in phase. The example in figure 9 displays
excellent force control on Celgard®, at a scan rate of 10 lines
per second, consistent with a gain in imaging bandwidth of
10–20x when compared to a standard AFM.

One question frequently asked is the effect of fast scanningon tip life. Tip abrasion and tip life are a concern when
covering more sample surface in a shorter amount of time.
A tip life limited to a few images would limit productivity anddrive up operating cost. Long tip life is also an indicator of
excellent, consistent force control.
Figure 8 shows an image of a TipCheck (Aurora
Nanodevices, Inc.) sample, a test structure for assessing tip
sharpness, with many sharp edges that make it useful forthe characterization and reconstruction of AFM tip shape.
These samples are known as “tip-eaters,” i.e., as causing
excessive tip wear, which sometimes limits their usefulness
for characterizing tips.
The sample was scanned at a rate of 17 lines per second.
The plot on the right measures the estimated tip radius,
reconstructed from the sample over successive images,
using the Blind Tip Estimation feature in the NanoScope
software. The graph illustrates that the fast imaging of avery rough, hard sample will wear the tip radius very slowly,(0.9 nanometers over 200 frames). This is a worst-case

Figure 9: Celgard ® oriented Polypropylene membrane(TappingMode, Broadband-A Probe) is a good indicator of AFM force
control. Tracking error would cause lateral forces to separate anddamage the smaller strands.

7

Application Examples for Survey, Screening and
Dynamics
As mentioned above, the three application areas that benefit
most from a higher-bandwidth AFM with the same data
quality, force control, operating cost and ease of use as astandard AFM can be categorized under survey, screening,and dynamics. The Dimension FastScan was applied toeach of these categories.
SurveySurvey is the understanding of the representative
morphologies of a heterogeneous, unknown sample. Thisis a very common situation when using an AFM (or any
microscopy technique) on a new sample. Especially forcomplex (e.g., biomaterial) samples, the large majorityof imaging time is often spent looking at enough sample
surface to understand what is important, rather thancapturing the final images that represent the sample.
The famous parable of the three blind men describing
an elephant comes to mind, one describing the tail, one
describing the trunk, and one describing the tusks. Coveringa larger area of the sample, with sufficient detail and within
an acceptable amount of time leads to a better, morebalanced view of the parts and their respective roles.
A higher bandwidth AFM can be applied toward this goalin different ways. On a rough sample, more sites can be
engaged and imaged in a shorter amount of time. The

NanoScope software’s MIRO image overlay capability canbe used to keep track of all the scans within one context,
and in relation to an overview optical image. On a fairly flat
sample, another way to survey the sample is to capture avery large scan area with very high pixel resolution. The
data can then be zoomed into and analyzed (even without
using further tool time) and representative areas can bemagnified and published. A big advantage of this method
is that one can decide on the best scale and framing after
taking all the data. Faster imaging brings the time for a highpixel resolution image (e.g., 16 megapixels) down fromseveral hours to a few minutes. The data shown in figure
10 consists of one 16-megapixel image of a 20-micron
scan range on a PTFE polymer film, acquired in 8 minutes,
with data zooms of various interesting morphologies, aswell as phase data for two of them. Looking at the overall
dataset, one has good confidence of the morphologies thatcan occur on this surface, while also achieving high-quality
images from the high-resolution dataset.
Screening
In screening applications, the space of possible phenomena
is typically well understood. However, the dependency
between an input parameter (such as an ingredientconcentration), or a process parameter, and a nanoscalemorphology or property (roughness, number of phases,
domain size, defect rate, or mechanical property) must be
understood and quantified. For this, it is necessary to image

Figure 10: 20µm, 16MPixel image of PTFE polymer film (left), acquired in 8 minutes. Right: Multiple data zooms showing detail and phasedata. Surveying a sample means to explore and understand it’s representative morphologies, and document them in publication quality
images. On sufficiently flat samples, one survey method is to take a large, high-resolution scan that can be explored offline for representative
morphologies, which can then be magnified and published.
8

(incl. Stage Navigation, Engage, Capture, Withdraw). 12 Samples, 60 Sites/Engages. Automated.

Figure 11: Screen of twelve amorphous drug formulation candidates (fractured film, 3µm scans, five sites per candidate). Batch analysis
shows material specific roughness with tight error bars; excipients with API load are smoother than blanks. This type screen is used
to verify compound compatibility, and to rapidly predict stability/shelf life, after brief stress aging. (Samples courtesy of M.E. Lauer,
F. Hoffmann-La Roche, Basel, Switzerland.)(2100 Frames were captured at 1 frame per second)

Figure 12: DNA loosely bound to mica treated by APS-method. TappingMode in buffer solution. Probe: Broadband-C. 1 frame/s. Shown are
3 of 2100 frames, showing the diffusion of the DNA over 35 minutes. This study of sample dynamics demonstrates 1 frame/s imaging, withthe typical, project-specific trade-off of frame rate and image quality. Good tracking must be maintained to minimize tip impact on the looselybound, fragile sample. (Sample courtesy of Y. Lyubchenko, Univ. of Nebraska Medical Ctr., USA.)

multiple sites on multiple samples, and to efficiently analyze
and quantify the morphology or property. Imaging speed isone element to throughput here, while multi-sample loading
and automation, fast engage, consistent operation without
user intervention, and data management and batch image
analysis are equally important elements.
Figure 11 shows an example of AFM screening fromthe pharmaceutical industry.8 In this case, the active
pharmaceutical ingredient (API) is combined (formulated)with an (inactive) excipient to form an amorphous solid,
with the goal of maximizing the API’s solubility afteringestion. The amorphous formulation is solid (frozen) atroom temperature but would otherwise phase separate.
To observe the possible phase separation with bulk

techniques, relatively macroscopic (~100nm) separation andre-crystallization of the API must first occur. The Dimension
FastScan AFM can catch the indicators of instability on
a much smaller size scale, much earlier. For routine use,
the AFM needs to give data on multiple sites of 100 or
more samples per day. This throughput is enabled by the
combination of >10x faster scan speeds, fast engage,
multi-sample automation, robustness of measurement, andultimate data quality.
Dynamics
The “classic” discipline for high-speed AFM is the timeresolved study of dynamic processes on the scale ofproteins and DNA. This application has driven much of the
9The Future of Rapid AFM Imaging
The idealistic notion of faster AFM imaging is almost as old
as the AFM itself. A number of implementations for specificapplications have demonstrated that great increases in AFM
imaging speed are possible. We have approached higher
speed AFM not as a certain set of applications, by certain
fields of research and on certain samples, but with thebelief that one would rather always image faster, however
not at the expense of quality, sample size or delicacy,
usability, or operating cost. That said, we do expect that a

References
1. G. Binnig, C.F. Quate, and Ch. Gerber, “Atomic Force Microscope,” Phys.
Rev. Lett., 56:930, (1986).
2. T. Ando, T. Uchihashi, N. Kodera, D. Yamamoto, A. Miyagi, M. Taniguchi,
and H. Yamashita, “High-Speed AFM and Nano-Visualization of Biomolecular
Processes,” Pflugers Arch - Eur J Physiol 456:211–25, (2008 Review).
3. J.H. Kindt, G.E. Fantner, J.A. Cutroni, and P.K. Hansma, “Rigid Designof Fast Scanning Probe Microscopes Using Finite Element Analysis,”
Ultramicroscopy 100 259–65, (August 2004).
4. A.D.L. Humphris, M.J. Miles, and J.K. Hobbs, “A Mechanical Microscope:High Speed Atomic Force Microscopy,”. Appl. Phys. Lett., 86, 034106-3(2005).
5. M.B. Viani, L.I. Pietrasanta, J.B. Thompson, A. Chand, I.C. Gebeshuber,
J.H. Kindt, M. Richter, H.G. Hansma, and P.K. Hansma, “Probing ProteinProtein Interactions in Real Time,” Nature Structural Biology 7 (8): 644-47,(August 2000).
6. J.H. Kindt, G.E. Fantner, and P.K. Hansma, “Scanner for Probe
Microscopy,” US Pat.No.7,278,298 B2 (2007).
7. L.A. Nagahara et al., “Mica Etch Pits as a Height Calibration Source for
Atomic Force Microscopy,” J Vas. Sci. Technol. B, 12/3, (1994).
8. M.E. Lauer, O. Grassmann, M. Siam, J. Tardio, L. Jacob, S. Page, J. Kindt,
A. Engel, and J. Alsenz, “Atomic Force Microscopy-Based Screening ofDrug-Excipient Miscibility and Stability of Solid Dispersions,” Pharmaceutical
Research (Nov. 2010).
9. L.S. Shlyakhtenko, A.A. Gall, A. Filonov, Z. Cerovac, A. Lushnikov, and Y.L.
Lyubchenko, “Silatrane-Based Surface Chemistry for Immobilization of DNA,
Protein-DNA Complexes and Other Biological Materials,” Ultramicroscopy 97,
279-87 (2003).
10. Y.L. Lyubchenko, “DNA Structure and Dynamics: An Atomic ForceMicroscopy Study,” Cell Biochem Biophys 41, 75-98 (2004).
11. J.L. Gilmore, Y. Suzuki, G. Tamulaitis, V. Siksnys, K. Takeyasu, and
Y.L. Lyubchenko, “Single-Molecule Dynamics of the DNA-EcoRII Protein
Complexes Revealed with High-Speed Atomic Force Microscopy,”
Biochemistry 2009, 48, 10492 -10498.
12. http://www.bruker-axs.com/dimension_fastscan_atomic_force_
microscope.html.

Authors
Johannes H. Kindt, Nghi Phan, Bede Pittenger, Adam Mednick, Andrea Slade,
Lin Huang, Weije Wang, Natalia Erina, James E. Shaw, and Stephen C. Minne(Bruker Nano Surfaces Division)

Acknowledgements
We would like to thank Dr. Matthias E. Lauer (F. Hoffmann-La Roche,
Basel) for the sample set of Amorphous Formulations, and Professor Yuri
Lyubchenko (Univ. Nebraska Medical Center) for the DNA and APS-Micasample system.

Bruker Nano Surfaces Divison
Santa Barbara, CA · USA+1.805.967.1400/800.873.9750
productinfo@bruker-nano.com

10

www.bruker.com/FastScan

Dimension, FastScan, Icon, Broadband, NanoScope, and TappingMode are trademarks or registered trademarks of Bruker Corporation. AN134, Rev. A0

Figure 12 shows three frames from a time sequence of2100 frames, captured at a rate of 1 frame per second, of
DNA in buffer solution, loosely bound to and diffusing onan APS-treated mica substrate.9-11 In the movie, different
motions of the DNA can be seen, including a “sliding”
motion of the DNA along its contour, and approximately
perpendicular to the scan direction.12 This indicates that the
DNA’s binding to the substrate is loose enough to allow it
to move, and diffusion is not dominated by the back-andforth scan motion of the AFM tip. This should lay a goodfoundation for the observation of more complex sample
systems, such as DNA-protein complexes, ATP-driven
systems, etc.

faster AFM will open up new areas of investigation over thefull range of applications, from routine industrial to molecularbiophysics. Most importantly, it will also allow researchers
to quickly and efficiently look at and understand a sample
at the nanoscale, using the breadth and content richness of
the AFM technique.

©2011 Bruker Corporation. All rights reserved. Clegard is a registered trademark of Celgard LLC, and is not associated with Bruker Nano Surfaces Division.

initial understanding of how to make AFMs faster, while
maintaining non-destructive tip-sample forces. What was
found is that the main bottle neck was cantilever dynamics,
and that to make cantilevers faster without sacrificing force
control, cantilevers must be made smaller. As soon as thatis done, a range of further requirements arise; to enable theuse of smaller cantilevers, to scan faster, and to capture
data faster. In this hunt for speed, it was found that theachievable frame rate scales roughly with the dimensionsof the cantilevers. It also scales with the data quality, with
the number of lines, and with the acceptable pixel blurcaused by loose tracking (parachuting). Achieving frame
rates >1frame per second is typically achieved by increasing
imaging bandwidth, and by trading image quality for speed.
For movie data, this can be quite acceptable to the humaneye. In an analogy, a frozen TV picture typically doesn’tlook that great individually, but the movie looks good. For
the FastScan to be more than a single-purpose movie
machine, it was important to have full AFM performanceat increased bandwidth, but to be able to further trade off
resolution for speed in the way of other high-speed AFMs,
and to maintain excellent control of tip-sample forces at highscan rates.