PFX2000 Series Basic Package

Battery Tester

PFX2000 Series Basic Package
Best suited for the evaluation on secondary batteries
5 V/5 A, 25 W × 2 channels(PFX2011 Basic Package)
20 V/10 A, 200 W × 1 channel(PFX2021 Basic Package)

PointPoint

2

Dependable
safety!

1All-in-one package!
This all-in-one package includes thenecessary and convenient application
software, load cable with alligator clips
for connecting to the test material andeverything you need to begin.

Equips various protections suchas OVP, UVP, OHP, OTP, etc toprevent the batteries from being
damaged by a system mulfunction
or operation mistake.

Point

3

High cost performance!
Realizes high-accuracy and highstability testing for 1ch and 2ch battery
tests at an affordable price.
application software

single-unit frame

Control unit

Charging/discharging power unit PFX2011
Fully-independent 2CH (5 V/5 A, 25 W)*This photo shows an example of the PFX2011 package.
The PC is not included.

Battery Tester
Basic Package

PFX2000
SERIES

Basic Package<Lineup>● PFX2011 Basic Package [5 V-5 A/ 2ch]
● PFX2021 Basic Package [20 V-10 A/ 1ch]
<Package contents>■ Charging/discharging power unit (PFX2011 or 2021) ■ Control unit ■ Unique single-unit frame■ Unique application software ■ Load cable for test material connection (with alligator clips)* PC is not included. The specifications of the unique application software that is provided with this product (BPChecker2000
BASIC Edition) are limited to 2-channel operation. The impedance measurement unit cannot be connected.
The other specifications are all the same as the BPChecker2000 FULL Edition application software (SD002).

Application Software
Examples ofapplications

Application software, BPChecker2000 provides centralized management includingsetting of test conditions, test execution and analysis of results.
Variousbatteries

Cellulartelephones

Digitalcameras

With the PFX2000 Series Basic Package and a Windows PC, you
can begin battery testing including PASS/NG tests, lifetime diagnosis (deterioration tests) and comparison tests. The PFX2000 Series
is a high-performance battery testing system that is used by battery
manufacturers. This package is all that is needed to perform highaccuracy, high-stability testing that meets the strict needs of battery
manufacturers.PFX2011 is suitable for characteristic evaluation for
single cell batteries and mobile phones.
PFX2021 is ideal for characteristic evaluation for laptop PC, digitalcameras, etc.

2It also allows external control of a thermostatic chamber (product of Espec Corp.)
via GPIB or RS232C communications, and it is capable of synchronized test withthe chamber temperature.
Recommended operating environment : CPU: Pentium IV 1 GHz or higher / Memory: Minimum 512 MB / Windows 2000
(SP4 + Update Rollup1), XP (SP2 or later, x86), Vista (x86, x64) / USB interface (For thermostatic chamber control,
GPIB or RS232C is also required.)▲ Example of screen display: The charging (discharging, charging + discharging) curve canbe overlaid on the display. The average, standard deviation, maximum value and minimum
value for the overlaid data can also be calculated for data analysis.

The entire operation can be managed by the application software (standard accessory)
The “BPChecker 2000 Basic Edition”, a standard accessory,
can manage the entire operation from the setting of thetest conditions, the execution of the test, and analyzing thetest result files. This software can controll the thermostaticchambers (manufactured by ESPEC) and also applies to thesynchronized test with the thermostatic chambers.
The recommended operating environment : CPU: Pentium
IV 1GHz or higher / Memory 512 MB or more / Windows 2000
Professional (SP4 + Update Roll up1), Windows XP (SP2 or laterwith Intel x86) or Windows Vista (Intel x86,x64) / USB interface(GPIB or RS232C inter face is required for controlling thethermostatic chamber)

Test Executive

Graph Viewer

This application controls the execution of the test. It starts and stops the test
and monitors the test execution. It provides a real-time graphical representation of the per-channel charging/discharging trends.

This application offers graphical representations of the charging/discharging
data for each cycle. It can display up to 99 sets of data overlaid one anotherin a single graph and perform statistical processing.

▲The figure shows the overlapped graph of charging curve (discharge, charge+ discharge), it is also capable of calculating the average, standard deviation,
max or min value, and the data analysis.

Conceptual Diagrams of Charging Mode Operation
CC-CV (constant current-constant voltage)

Conceptual Diagrams of Discharging Mode Operation
CP (constant power)

CC (constant current)

Charging time

Discharging time

Discharging time

CV timeBatteryvoltage

CV voltage

It current

Charging
current

Batteryvoltage
Dwell time

It time[Termination conditions] Time, CV time, current,
and temperature

CC (constant current)

Charging
current

Final voltage
Dwell time
Discharging current[Termination conditions] Time and voltage

CC pulse (constant current 8-value/20-value pulse)*

Batteryvoltage

Final voltage

Charging
current

Limit current[Termination conditions] Time and voltage

CP pulse (constant power 20-value pulse) * For PFX2021 only
Discharging timeDischarging time
Charging time

Charging
current

Dwell time

[Termination conditions]  Time, voltage, –ΔV,
temperature, and ΔT/Δt

CC PWM (constant current PWM pulse)
PWM time

Maximumvoltage

Maximumvoltage
Pulse time 1

Minimum voltage

Limit current

Pulse
discharging
current[Termination conditions]Time and low voltage* The above diagram applies to
the 8-value pulse of PFX2011.
The 20-value pulse issupported only for PFX2021.

Charging time

Dwell timeHigh voltage
Batteryvoltage
Pulse
discharging
current

Low voltage

Pulse
discharging
current[Termination conditions]Time and low voltage

Dwell timeHigh voltage

Low voltage

Maximumvoltage

On timeIt time

Pulse time 2Charging
current

Final voltage

Final voltage

Batteryvoltage

Batteryvoltage

Batteryvoltage

Batteryvoltage–dV

Maximum voltage

Dwell time

Constant
current
Pulse current 1 Pulse current 2

CC interval [1]CC pulse interval [2]Pulse on current

Pulse off current

Dwell time

PWM interval [3][Termination conditions] Time and off time3

Function specifications
PFX2011

Constantcurrent/constant voltage (CC-CV), Constantcurrent (CC)Pulse

PWM pulse (CC-PWM)
Constant current (CC), Constant power (CP)Constant pulse current (CC Pulse)–RangePulse
constantpower
discharge

Discharge function

Pulse

Battery voltage, Charge/discharge current, Battery temperature, Capacity, TimeTime width Accuracy*10

Pulse

Battery voltage (PeakPoint, Multi Point), Charge/discharge current,
Battery temperature, Capacity, Time

Range

1 mA
±(0.15 %+3.0 mA)

CD board overheat (OHP)

Response*6 *1050 μs (typical)

70 μs (typical)

CD board error (CD Alm)

Range*7

Watchdog timer

10 μs

Measurement accuracy

AC power line error (AC off)
Alarm monitoring

Electrical specifications
PFX2011

PFX2021

Rated output
Number of outputs

2 ch

1 ch

Charge current range

0.0 mA to 5000.0 mA(High range)
0.00 mA to 500.00 mA(Low range)

0 mA to 10000 mA

Charge voltage range

0.0000 V to 5.0000 V

0.000 V to 20.000 V

Discharge current range

0.0 mA to 5000.0 mA(High range)
0.00 mA to 500.00 mA(Low range)

0 mA to 10000 mA

Discharge voltage range-0.5000 V to 5.0000 V-2.000 V to 20.000 V

25.00 W

200.00 W

Staticcurrent
measurement
Staticvoltagemeasurement

Pulse
charge/discharge
current

Range

0.0 mA to 5000.0 mA(High range)
0.00 mA to 500.00 mA(Low range)

0 mA to 10000 mA

Accuracy*1 *10±(0.05 %+1.0 mA)(High range)±(0.05 %+0.10 mA)(Low range)±(0.15 %+2.0 mA)

Resolution

0.1 mA (High range)
0.01 mA (Low range)

1 mA

Ripple*2 *10

1 mA rms (High/Low range)

3 mA rms

Range

0.0000 V to 5.0000 V

0.000 V to 20.000 V

Accuracy*3 *10±(0.03 %+1.0 mV)±(0.10 %+3.0 mV)

Resolution

0.1 mV

1 mV

2 mV rms

5 mV rms

Range

0.01 W to 25.00 W (High range)
0.001 W to 2.500 W (Low range)

0.02 W to 200.00 W

Accuracy*4 *10±(0.10 %+10.0 mW)(High range)±(0.10 %+2.0 mW) (Low range)±(0.50 %+20.0 mW)

10 mW (High range)
1 mW (Low range)

10 mW

0.0 mA to 5000.0 mA(High range)
0.00 mA to 500.00 mA(Low range)

0 mA to 10000 mA

Resolution

0.1 mA (High range)
0.01 mA (Low range)

1 mAAccuracy*1 *10±(0.07 %+1.0 mA) (High range)±(0.07 %+0.10 mA) (Low range)±(0.15 %+3 mΑ)

Number ofsettings

8 values

20 values

Response*6 *1050 μs (typical)

70 μs (typical)
0.50 ms to 65000.00 ms±(0.05 %+0.05 ms)
10 μs

Range

0.0 mA to 5000.0 mA(High range)
0.00 mA to 500.00 mA(Low range)

0.0 mA to 10000.0 mA

Accuracy*10 *11±(0.04 %+0.8 mA) (High range)±(0.04 %+0.08 mA) (Low range)±(0.15 %+1.5 mA)

Resolution

0.1 mA (High range)
0.01 mA (Low range)

0.1 mA

Range-0.5000 V to 5.0000 V-2.0000 V to 20.0000 V

Accuracy*10 *11±(0.02 %+1.0 mV)
0.1 mVRange

0.0 mA to 5000.0 mA(High range)
0.00 mA to 500.00 mA(Low range)

0.0 mA to 10000.0 mA

Accuracy*10±(0.10 %+1.0 mA) (High range)±(0.10 %+0.10 mA) (Low range)±(0.20 %+3.0 mA)

Resolution

0.1 mA (High range)
0.01 mA (Low range)

0.1 mA
Average current

Range
Pulse
battery
voltage±(0.10 %+2.0 mV)

Resolution

Measuredvalue*12

Accuracy of settings

Resolution±(0.05 %+0.05 ms)

Resolution

Communication error

Time width Accuracy*10

0.50 ms to 65000.00 ms

Time width Accuracy*10

DUT (battery) connection error (Connection Error)

Range*710 μs±(0.07 %+1.0 mA) (High range)±(0.07 %+0.10 mA) (Low range)

Pulse
PWM
chargePS board error (PS Alm)

Pulse
constant
current
discharge

0 mA to 10000 mA

Accuracy*1 *10

DUT overtemperature protection (OTP)

Range

0.0 mA to 5000.0 mA(High range)
0.00 mA to 500.00 mA(Low range)
0.1 mA (High range)
0.01 mA (Low range)

Overcharge capacity protection (OAH)

Resolution*5±(0.05 %+0.05 ms)

Resolution

Undervoltage (overdischarge) protection: Software UVP, Hardware UVPStaticconstant
power
discharge

5.00 ms to 65000.00 ms–

Resolution

Overvoltage (overcharge) protection: Software OVP, Hardware OVP

Ripple*2 *10

20 values
2 ms (typical)*9Range

Protection function

Staticconstant
voltagecharge±(0.5 %+20.0 mW)*8–

Number ofsettings

StaticStaticconstant
current
charge/discharge10 mW

Accuracy*10

Update rate

Measurement function

0.02 W to 200.00 W

Resolution

Constant pulse power (CP Pulse)

Maximum charge/discharge power

PFX2021

Accuracy of settings

StaticStaticPFX2011

PFX2021

Charge function-0.5000 V to 5.0000 V

Accuracy*10-2.0000 V to 20.0000 V±(0.05 %+1.0 mV)±(0.15 %+2.0 mV)

Resolution

0.1 mV

Measurement point

High voltage, Low voltage, Arbitrary

General Specifications
PFX2011
Input power
Powerconsumption

PFX2021
AC 100 V 50/60 Hz

At ratedoutput
With noload

External dimensions(Approx.)Weight400 VAmax

800 VAmax

60 VAmax

50 VAmax

85.5 W x 177 H x 523 D mm (3.37 W x 3.97 H x 20.59 D inch)
85.5 W x 177 H x 560 D mm maximum (3.37 W x 3.97 H x 22.05 D inch maximum)
Approx. 4 kg (8.82 lbs)

Approx. 4.5 kg (9.92 lbs)

*1. With respect to the specified current within the rated range.
*2. Maximum value at 10 Hz to 500 kHz.
*3. With respect to the specified voltage within the rated range.
*4. With respect to the specified power at a battery voltage of 0.5 V or greater on the PFX2011, or 2 V or greateron thePFX2021.
*5. Voltage operation range of constant power discharge (warranted value) 0.5 V to 5 V on the PFX2011, 2 V to20 Von the PFX2021.
*6. At 10 % to 90 % of the pulse current waveform when rated current is set. Short-circuit at the tip of the 7-mloadcable.
*7. The pulse time width is measured at the mesial point of the pulse.
*8. With respect to the specified power at a battery voltage of 2 V or greater.
*9. Indicates the update rate of the control current by software computation. It is always a fixed time regardlessof thepulse time width.
*10. Ambient temperature: 18 °C to 28 °C*11. With respect to the actual value within the rated range.
*12. Measures the average current every 500 ms.

●Distributor/Representative

Southwood 4F,6-1 Chigasaki-chuo,Tsuzuki-ku,Yokohama,224-0032,Japan
Phone: (+81)45-482-6353,Facsimile: (+81)45-482-6261,www.kikusui.co.jp

1-310-214-0000 www.kikusuiamerica.com
3625 Del Amo Blvd, Suite 160, Torrance, CA 90503
Phone : 310-214-0000 Facsimile : 310-214-0014

www.kikusui.cn
Room 305,Shenggao Building , No.137,Xianxia Road, Shanghai City, China
Phone : 021-5887-9067 Facsimile : 021-5887-9069

Printed in Japan■ All products contained in this catalogue are equipment and devices that are premised on use under the supervision ofqualified personnel, and are not designed or produced for home-use or use by general consumers. ■ Specifications, designand so forth are subject to change without prior notice to improve the quality. ■ Product names and prices are subjectto change and production may be discontinued when necessary. ■ Product names, company names and brand names
contained in this catalogue represent the respective registered trade name or trade mark. ■ Colors, textures and so forthof photographs shown in this catalogue may differ from actual products due to a limited fidelity in printing. ■ Although every
effort has been made to provide the information as accurate as possible for this catalogue, certain details have unavoidably
been omitted due to limitations in space. ■ If you find any misprints or errors in this catalogue, it would be appreciated if
you would inform us. ■ Please contact our distributors to confirm specifications, price, accessories or anything that may beunclear when placing an order or concluding a purchasing agreement.

Issue:Oct.2019 201910PDFEC21a