TS-42531F-CE-v02

English㻌 02-2023-Histofine®

DAB-2V
Peroxidase Chromogen/Substrate solution

NICHIREI BIOSCIENCES INC.
6-19-20, Tsukiji, Chuo-ku, Tokyo 104-8402, JAPAN
Phone: 81-3-3248-2208 Facsimile: 81-3-3248-2243

3,3’-DIAMINOBENZIDINE TETRAHYDROCHLORIDE (DAB) SUBSTRATE SOLUTION
SUBSTRATE FOR HORSERADISH PEROXIDASE
Package/Code㻌 㻌 500 tests/425312F, 1500 tests/425314F

I1. DESCRIPTION-Histofine® DAB-2V is designed as one of reagents to visualize antigen binding primary antibody on tissue sections.

I-2. INTENDED USE-Histofine® DAB -2V is designed to allow immunohistochemical tests, to reveal antigens that reacts with a user-supplied primary on
tissues and cells. The product is characterized by a coloring reagent/substrate for coloring a detection system reagent labeled withperoxidase. The product is one of the specific chromogen/substrate solutions for peroxidase and develops Erown colored precipitate byreaction with peroxidase. The result provides a physiological or pathological state.
For In-vitro Diagnostic Use. For manual staining. Staining results are qualitative. For formalin-fixed paraffin-embedded human tissue
sections.
For professional users.
Interpretation must be made within the context of the patient’s clinical history and other diagnostic tests by a qualified pathologist.

I-3. PRINCIPLE
The antigen / antibody / Universal Immuno-peroxidase Polymer complex can be prepared by allowing the reagent to react with a primary
antibody bound to the antigen on tissue section. The enzymatic activity of this complex results in a colored deposit, thus staining theantigen site.

II. COMPONENTS
1. Reagents provided[500 tests]
Reagent A: DAB solution concentrate (Liquid)…
1.2 ml×2
3,3’-diaminobenzidine tetrahydrochloride (DAB) chromogen solution
Reagent B: DAB buffer (Liquid)…
30 ml×2
Hydrogen peroxide
2. Accessories
Tube
…
×1
Chip (For Reagent B, colored)…
×1
Dropping bottle…
×1[1500 tests]
1.2 ml×6
30 ml×6×1
×1
×1

III. USAGE
1. Preparation of DAB solution
Add 1 drop (approx. 20 μl) of Reagent A Chromogen reagent per 500 μl of Reagent B Chromogen substrate into graduated tube, andplace its cap then mix carefully without foaming.
Remove full dosage of the DAB solution from the tube to dropping bottle using pipette as well for use.
Store in light shielded condition at 2 to 8°C and use within 2 weeks after the preparation.
Note: 1. Use DAB solution removing foams or exclusive of foams when its surface is foamed by mixing.
2. Do not use mixture of DAB solution prepared different day.
2. Addition and Reaction of DAB solution
1. Apply DAB solution using dropping bottle covering the tissue section on the slide completely and incubate at room temperature for 5
to 20 minutes after addition, reaction and washing of peroxidase conjugate in the immuno- histochemical staining process.
2. After adequate chromogenic reaction, wash the slide well with tap water.
3. Mount the slide with permanent mounting media and fix with cover slip after dehydrated in graded series of alcohol and cleared inxylene.
Note: 1. Adjust the reaction time by microscopic observation.
2. At the same time, staining of negative control slide using such as PBS in place of primary antibody is recommended to beprepared for checking the staining result.

-Histofine is NICHIREI BIOSCIENCES’s registered trademark. The registration countries would be referred us.     TS-T-42531F-CE-E-v02

3. Accessories
Tube
For mixing of Reagent A and Reagent B. (Refer to aforementioned Preparation of DAB solution.)
Chip (For Reagent B, colored)
Reagent B is provided with translucent chip placed on its bottle. However the chip is not allowed for applying solution drop by drop,
therefore it is not suitable for small volume preparation. For such preparation, replace the translucent chip by colored chip provided
exclusively for dropping is recommended. The colored chip is suitable for applying solution drop by drop.
Dropping bottle
Dropping bottle is used for storage of reagents, not only for dropping of DAB solution.
Note: Tube and dropping bottle are reusable after wash with purified water and dry completely.

IV. STAINING PROCEDURE
Refer to the instruction* of-Histofine® Simple Stain MAX PO (MULTI) for staining procedures such as Principle, Materials required
but not supplied, Specimen preparation, Control slides, Interpretation of results and General limitation. Validation is required when otherdetection systems besides above are used.
*Download of the instruction is available at following website:
http://www.nichirei.co.jp/bio/english/products/pdf/CE_MAXPO_MULTI.pdf

V. STORAGE, HANDLING & SHELF LIFE
1. Store at 2-8°C.
2. Store prepared reagent, DAB solution, in light shielded condition at 2 to 8°C and use within 2 weeks after the preparation.
3. Confirm expiration date indicated on the labels of vial and carton before use.
4. Adjust this product to room temperature before use.
5. Refrigerate this product promptly after use.
6. The reagent is stable 18 months after manufacturing.

VI. PRECAUTIONS
1.
2.
3.
4.
5.
6.
7.
8.
9.
10.
11.
12.

Before using this reagent, read these instructions.
Do not use reagents after the expiration date.
For professional users.
Specimens, before and after fixation, and all other materials exposed to them, should be handled like biohazardous materials withproper precautions.
Inhalation or ingestion of the highly allergic fixative formaldehyde is harmful. Wear protective mask. If swallowed, induce vomiting.
If skin or eye contact occurs, wash thoroughly with water.
Organic reagents are flammable. Do not use near open flame.
Never pipette reagents by mouth and avoid their contact with skin, mucous membranes and clothes.
Avoid microbial contamination of reagents as incorrect result may occur.
Avoid splashing of reagents or generation of aerosols.
As a chromogen, DAB solution should be handled with care for it contains carcinogen䠊
Unused solution should be disposed of according to local, State and Federal regulations.
Any serious incident that has occurred in relation to-Histofine® DAB-2V shall be reported to the manufacturer and the competentauthority of the Member State in which the user and/or the patient is established.

VII. REFERENCE(1) Mokrỳ, et al. Versatility of immunohistochemical reactions: comprehensive survey of detection systems. Acta medica 1996 Apr
39:129-40(2) Yamada K, et al. In vitro assessment of antitumor immune responses using tumor antigen proteins produced by transgenic silkworms.
J Mater Sci Mater Med. 2021 May 17;32(6):58.

Catalog/Code Number

Temperature Limitations

In Vitro Diagnostic Medical
DeviceManufacturer

Batch CodeContains Sufficient for䚾 N䚿 Tests

Use ByConsult Instructions
for UseAuthorized Representativein the European Community

CE-mark, code of thenotified body

For IVD Performance
Evaluation only

SampleNICHIREI BIOSCIENCES INC.

6-19-20, Tsukiji, Chuo-ku, Tokyo 104-8402, JAPAN
Phone: 81-3-3248-2208, Facsimile: 81-3-3248-2243

MedEnvoy Global B.V.

Prinses Margrietplantsoen 33-Suite 1232595 AM The Hague
The Netherlands-Histofine is NICHIREI BIOSCIENCES’s registered trademark. The registration countries would be referred us.     TS-T-42531F-CE-E-v02

Français㻌 02-2023-Histofine®

DAB-2V
Solution de substrat/chromogène de la peroxydase

NICHIREI BIOSCIENCES INC.
6-19-20, Tsukiji, Chuo-ku, Tokyo 104-8402, JAPAN
Phone:81-3-3248-2208, Facsimile:81-3-3248-2243

SOLUTION DE SUBSTRAT 3,3'-DIAMINOBENZIDINE TÉTRACHLORHYDRATE (DAB)
SUBSTRAT POUR PEROXYDASE DE RAIFORT
Conditionnement/Code㻌 㻌 500 tests/425312F, 1 500 tests/425314F

I1. DESCRIPTION-Histofine® DAB-2V est l'un des réactifs conçus pour visualiser la fixation de l'anticorps primaire sur l'antigène dans des sections de tissu.

I-2. APPLICATIONS-Histofine® DAB-2V est conçu pour détecter des antigènes dans des coupes de tissus ou des cellules qui réagissent avec des anticorpsprêts à l'emploi par des techniques immunohistochimiques. Le produit est caractérisé par un réactif/substrat de coloration pour colorer unréactif de système de détection marqué à la peroxydase. Le produit est l'une des solutions de substrat/chromogène spécifiques pour laperoxydase et il développe un précipité brun par réaction avec la peroxydase. Le résultat fournit un état physiologique ou pathologique.
Pour usage diagnostic in vitroPour une coloration manuelle. Les résultats de la coloration sont qualitatifs.Pour les coupes de tissus fixés au
formol et inclus en paraffine. A usage professionnel exclusivement.
L’Interprétation des résultats doit être faite par un biologiste qualifié, en prenant en considération l’historique de la pathologie du patientainsi que le bilan complet des tests qui ont été pratiqués.

I-3. PRINCIPLE
Le Complexe de Polymère Universel antigène / anticorps / Immuno-peroxydase peut être formé en mettant en présence le réactif qui va réagir
avec l’anticorps primaire en se fixant sur l’antigène contenu dans la coupe de tissus. L’activité enzymatique de ce complexe fera apparaître
un dépôt coloré, mettant ainsi en évidence la présence du site antigénique recherché dans l’échantillon testé.

II. COMPOSANTS
1. Réactifs fournis[500 tests]
Réactif A : Concentré de solution DAB (liquide)…
1,2 ml×2
Solution chromogène 3,3'-diaminobenzidine tétrachlorhydrate (DAB)
Réactif B : Tampon DAB (liquide)…
30 ml×2
Peroxyde d'hydrogène
2. Accessoires
Tube
…
×1
Puce (pour réactif B, colorée)…
×1
Flacon compte-gouttes…
×1[1 500 tests]
1,2 ml×6
30 ml×6×1
×1
×1

III. UTILISATION
1. Préparation de la solution DAB
Ajouter 1 goutte (env. 20 μl) de réactif chromogène (réactif A) par 500 μl de substrat chromogène (réactif B) dans un tube gradué, placer son
bouchon puis mélanger soigneusement en évitant la formation de mousse.
Retirer du tube le dosage complet de la solution DAB dans le flacon compte-gouttes à l'aide d'une pipette.
Conserver à l'abri de la lumière entre 2 et 8 C et utiliser dans les 2 semaines suivant la préparation.
Remarque :1. Utiliser la solution DAB en prenant soin de retirer les mousses ou l'utiliser à l'exclusion des mousses qui se sont formées pendant lemélange.
2. Ne pas utiliser un mélange de solutions DAB préparées à des jours différents.

2. Ajout et réaction de la solution DAB
1. Appliquer la solution DAB à l'aide du flacon compte-gouttes en recouvrant entièrement la section de tissu sur la lame et incuber à latempérature ambiante pendant 5 à 20 minutes après l'ajout, la réaction et le lavage du conjugué peroxydase au cours du processus decoloration immunohistochimique.
2. Après une réaction chromogène adéquate, laver le puits de lame à l'eau du robinet.
3. Monter la lame avec un support de montage permanent et la fixer avec la lamelle après déshydratation en une série graduée d'alcool etlibération des xylènes.
Remarque :1. Régler le temps de réaction par observation microscopique.
2. Au même moment, il est recommandé de préparer une coloration de la lame de contrôle négatif en utilisant par exemple un tampon
phosphate salin (PBS) à la place de l'anticorps primaire afin de contrôler le résultat de la coloration.

-Histofine® est la marque déposée de : NICHIREI BIOSCIENCES’. Nous contacter pour obtenir la liste des pays où le produit est enregistré.

TS-42531F-CE-F-v02

3. Accessoires
Tube
Pour mélanger le réactif A et le réactif B. (Se reporter à la section susmentionnée Préparation de la solution DAB.)
Puce (pour réactif B, colorée)
Le réactif B est fourni avec une puce translucide placée sur son flacon. Cependant, la puce n'est pas conçue pour une application goutte àgoutte de la solution ; elle ne convient donc pas pour une préparation de petits volumes. Pour une telle préparation, remplacer la pucetranslucide par une puce colorée fournie exclusivement pour l'application goutte à goutte. La puce colorée convient pour l'application goutteà goutte de la solution.
Flacon compte-gouttes
Un flacon compte-gouttes est utilisé à la fois pour conserver les réactifs et pour appliquer la solution DAB au goutte à goutte.
Remarque : Le tube et le flacon compte-gouttes sont réutilisables après lavage à l'eau purifiée et séchage complet.

IV. PROCÉDURE DE COLORATION
Pour les procédures de coloration, se reporter aux instructions* de-Histofine® Simple Stain MAX PO (MULTI) concernant le principe,le matériel requis mais non fourni, la préparation de l'échantillon, les lames témoins, l'interprétation des résultats et la prescription générale.
Une validation est nécessaire si d'autres systèmes de détection que ceux susmentionnés sont utilisés.
*Les instructions peuvent être téléchargées sur le site Internet suivant :http://www.nichirei.co.jp/bio/english/products/pdf/CE_MAXPO_MULTI.pdf

V. CONSERVATION, MANIPULATION & PEREMPTION
1. Conserver entre 2 et 8 C.
2. Conserver le réactif préparé/la solution DAB à l'abri de la lumière entre 2 et 8 °C et les utiliser dans les 2 semaines suivant la préparation.
3. Vérifier la date d'expiration indiquée sur les étiquettes du flacon et du carton avant utilisation.
4. Amener ce produit à la température ambiante avant utilisation.
5. Réfrigérer ce produit rapidement après utilisation.
6. Le réactif est stable Durant 18 mois après sa date de fabrication.

VI. PRECAUTIONS D’EMPLOI
1. Lire soigneusement ce mode d’emploi avant de réaliser ce test.
2. Ne pas utiliser ce produit après la date d'expiration.
3. A usage professionnel exclusivement.
4. Avant et après fixation, manipuler les échantillons ainsi que toutes autres substances auxquelles ils sont exposés avec toutes les précautions
s’appliquant aux matières présentant un danger de contamination biologique.
5. Le fixateur à base de formol est sévèrement allergisant et présente un grave danger en cas d’inhalation ou d’ingestion. Porter un masquede protection. En cas d’ingestion, provoquer des vomissements. En cas de contact avec la peau ou les yeux, rincer abondamment avec del’eau.
6. Les réactifs organiques sont inflammables. Ne pas les utiliser en présence d’une flamme ouverte.
7. Ne jamais pipeter les réactifs à la bouche, éviter tout contact avec la peau, les muqueuses et les vêtements
8. Eviter toutes contaminations microbiennes des réactifs, car celles-ci peuvent être la cause de faux négatifs ou faux positifs.
9. Eviter de faire gicler les réactifs ou de générer des vapeurs.
10. La solution chromogène DAB doit être manipulée avec précaution car elle contient des éléments cancérigènes.
11. Eliminer les réactifs non utilisés selon les dispositions légales locales, départementales et nationales.
12. Tout incident grave survenu en rapport avec
-Histofine® DAB-2V doit être signalé au fabricant et à l'autorité compétente de l'État
membre dans lequel l'utilisateur et/ou le patient est établi.

VII. BIBLIOGRAPHIE(1) Mokrỳ, et al. Versatility of immunohistochemical reactions: comprehensive survey of detection systems. Acta medica 1996 Apr 39:12940(2) Yamada K, et al. In vitro assessment of antitumor immune responses using tumor antigen proteins produced by transgenic silkworms. J
Mater Sci Mater Med. 2021 May 17;32(6):58.

Catalog/Code Number

Temperature Limitations

In Vitro Diagnostic Medical
DeviceManufacturer

Batch CodeContains Sufficient for䚾 N䚿 Tests

Use ByConsult Instructions
for UseAuthorized Representativein the European Community

CE-mark, code of thenotified body

For IVD Performance
Evaluation only

SampleNICHIREI BIOSCIENCES INC.

6-19-20, Tsukiji, Chuo-ku, Tokyo 104-8402, JAPAN
Phone: 81-3-3248-2208, Facsimile: 81-3-3248-2243

MedEnvoy Global B.V.

Prinses Margrietplantsoen 33-Suite 1232595 AM The Hague
The Netherlands-Histofine® est la marque déposée de : NICHIREI BIOSCIENCES’. Nous contacter pour obtenir la liste des pays où le produit est enregistré.

TS-42531F-CE-F-v02

German㻌 02-2023-Histofine®

DAB-2V
Peroxidase-Chromogen/SubstratlösungNICHIREI BIOSCIENCES INC.
6-19-20, Tsukiji, Chuo-ku, Tokyo 104-8402, JAPAN
Phone:81-3-3248-2208, Facsimile:81-3-3248-2243

3,3'-DIAMINOBENZIDIN-TETRAHYDROCHLORID (DAB) SUBSTRATLÖSUNGSUBSTRAT FÜR MEERRETTICHPEROXIDASE
Verpackung/Code 500 Tests/425312F, 1500 Tests/425314F

I1. BESCHREIBUNG-Histofine® DAB-2V ist als eines der Reagenzien konzipiert worden, um den primären Antigenbindungsantikörper in Gewebeschnitten
sichtbar zu machen.

I-2. VERWENDUNGSZWECK-Histofine® DAB -2V wurde zur Durchführung immunhistochemischer Tests entwickelt. Das System weist auf Geweben und Zellen
primärantikörper nach, die vom Anwender zum Erkennen von Antigenen eingesetzt werden.Das Produkt ist gekennzeichnet durch ein
färbendes Reagenz/Substrat zum Färben eines mit Peroxidase markierten Detektionssystemreagenzes. Das Produkt ist eine derspezifischen Chromogen-/Substratlösungen für Peroxidase und entwickelt durch Reaktion mit Peroxidase eine braun gefärbte Ablagerung.
Das Ergebnis liefert einen physiologischen oder pathologischen Zustand.
Zur Anwendung in der In-Vitro-Diagnostik. Für manuelle Färbung. Die Färbeergebnisse sind qualitativ. Für formalinfixierte, in Paraffin
eingebettete Gewebeschnitte. Für professionelle Anwender.
Die Interpretation der Ergebnisse sollte stets im Zusammenhang mit dem klinischen Bild der Patienten und weiteren diagnostischen
Parametern durch einen qualifizierten Pathologen erfolgen.

I-3. TESTPRINZIP
Der Antigen/Antikörper/Universelle Immun-Peroxidase Polymerkom- plex kann hergestellt werden, in dem das Reagenz mit einem
primärantikörper reagiert, der an ein Antigen auf einem Gewebeschnitt gebunden ist. Die enzymatische Aktivität dieses Komplexes
resultiert in einem farbigen Niederschlag, der die Stelle des nachzuweisenden Antigens färbt.

II. KOMPONENTEN
1. Zur Verfügung gestellte Reagenzien[500 Tests]
Reagenz A: DAB-Lösungskonzentrat (flüssig)…
1,2 ml × 2
3,3'-Diaminobenzidin-Tetrahydrochlorid (DAB) -Chromogen-Lösung
Reagenz B: DAB-Puffer (flüssig)…
30 ml × 2
Wasserstoffperoxid
2. Zubehör
Röhrchen…
×1
Chip (für Reagenz B, farbig)…
×1
Tropfflasche…
×1[1500 Tests]
1,2 ml × 6
30 ml × 6
×1
×1
×1

III. VERWENDUNG
1. Herstellung der DAB-Lösung
Dem Messröhrchen 1 Tropfen (ca. 20 μl) Reagenz-A-Chromogen-Reagenz pro 500 μl Reagenz-B-Chromogen-Substrat zugeben, den
Deckel aufsetzen und dann vorsichtig unter Vermeidung von Schaumbildung mischen.
Die volle Dosis der DAB-Lösung mithilfe einer Pipette aus dem Röhrchen der Tropfflasche nehmen.
Das Produkt bei 2 - 8 °C lagern und schützen und innerhalb von 2 Wochen nach Zubereitung einsetzen.
Hinweis:1. DAB-Lösung zum Entfernen von Schaum oder Schaumausschluss, verwenden, wenn die Oberfläche durch Vermischung Schaum
aufweist.
2. Keine DAB-Lösung, die an einem anderen Tag zubereitet wurde, benutzen.
2. Zugabe und Reaktion von DAB-Lösung
1. Die DAB-Lösung mit einer Tropfflasche auftragen, bis der Gewebeabschnitt auf dem Objektträger vollständig bedeckt ist, undanschließend während 5 - 20 Minuten bei Raumtemperatur, nachdem Hinzufügen, Reagieren und Abwaschen des PeroxidaseKonjugats im immunhistochemischen Färbungsprozess, inkubieren lassen.
2. Den Objektträger nach ausreichender chromogener Reaktion gründlich mit Leitungswasser abwaschen.
3. Den Objektträger mit einem permanenten Eindeckmittel eindecken und mit dem Deckglas fixieren, nachdem er in abgestuften
Alkoholreihen entwässert und in Xylol geklärt wurde.
Hinweis:1. Die Reaktionszeit durch mikroskopische Beobachtung anpassen.
2. Gleichzeitig wird empfohlen, die Färbung eines Negativkontrollobjektträgers unter Verwendung von PBS anstelle von primärem
Antikörpern vorzubereiten, um das Färbeergebnis zu überprüfen.
-Histofine ist ein geschütztes Warenzeichen von NICHIREI BIOSCIENCES. Die Registrierungsländer würden       TS-42531F-CE-G-v02

3. Zubehör
Tube
Zum Mischen von Reagenz A und Reagenz B. (Siehe vorstehende Vorbereitung der DAB-Lösung.)
Chip (für Reagenz B, gefärbt)Reagenz B wird mit durchscheinendem Chip auf der Flasche geliefert. Der Chip darf jedoch nicht tropfenweise aufgetragen werden, er
eignet sich daher nicht für die Vorbereitung kleiner Mengen. Für eine solche Vorbereitung wird empfohlen, den durchscheinenden Chip
durch einen farbigen Chip zu ersetzen, der ausschließlich zum Tropfen bereitgestellt wird. Der farbige Chip eignet sich zum tropfenweisen
Zuführen der Lösung.
Tropfflasche
Tropfflaschen werden zur Lagerung von Reagenzien verwendet und nicht für den Tropfvorgang von DAB-Lösung.
Hinweis: Röhrchen und Tropfflasche können nach dem Waschen mit destilliertem Wasser wiederverwendet und vollständig getrocknet
werden.

IV. FÄRBEVERFAHREN
Siehe Anweisung* über-Histofine® MAX PO (MULTI) Einfach-Färbung hinsichtlich Richtlinien, erforderliche Materialien, die nicht
im Lieferumfang enthalten sind, Probenvorbereitung, Kontrollobjektträger, Auswertung von Ergebnissen und allgemeine
Einschränkungen bei Färbeverfahren. Eine Validierung ist erforderlich, wenn andere Erkennungssysteme, als die oben genannten,
verwendet werden.
* Der Download der Anleitung ist auf folgender Website verfügbar:
http://www.nichirei.co.jp/bio/deutsch/products/pdf/CE_MAXPO_MULTI.pdf

V. LAGERUNG, HANDHABUNG UND HALTBARKEIT
1. Bei 2 - 8 °C lagern.
2. Das vorbereitete Reagenz und die DAB-Lösung, lichtgeschützt bei 2 - 8 °C aufbewahren und innerhalb von 2 Wochen nach Zubereitung
einsetzen.
3. Vor dem Gebrauch prüfen, ob das Verfallsdatum auf den Etiketten der Durchstechflasche und des Kartons angegeben ist.
4. Das Produkt vor Gebrauch auf Raumtemperatur erwärmen.
5. Das Produkt nach Gebrauch sofort kühlen.
6. Nach Herstellung ist das Reagenz 18 Monate stabil.

VI. VORSICHTSMASSNAHMEN
1.
2.
3.
4.

Vor Verwendung des Reagenzes bitte die Gebrauchsanweisung lesen.
Verfallene Reagenzien dürfen nicht mehr benutzt werden.
Nur für professionelle Anwender.
Das Reagenz enthält Material tierischen Ursprungs. Es kann nicht ausgeschlossen werden, dass auf Grund der Herstellungsverfahren
auch Spuren von Material menschlichen Ursprungs enthalten sind. Wie alle Produkte biologischen Ursprungs muss auch dieses
entsprechend gehandhabt werden.
5. Einatmen oder Verschlucken des hochallergischen Formaldehyds ist gesundheitsschädlich. Schutzmaske tragen. Bei Verschlucken
Erbrechen induzieren. Bei Haut- oder Augenkontakt gut mit Wasser auswaschen.
6. Organische Reagenzien sind entzündlich, bitte nicht in der Nähe einer offenen Flamme verwenden.
7. Die Reagenzien nie mit dem Mund pippetieren und den Kontakt mit Haut, Schleimhäuten und Kleidung vermeiden.
8. Mikrobielle Kontamination der Reagenzien vermeiden, weil diese die Ergebnisse verfälschen könnten.
9. Das Verspritzen von Reagenzien oder das Entstehen von Aerosolen
ist zu vermeiden.
10. Die chromogene DAB-Lösung sollte mit Vorsicht behandelt werden, weil sie Karzinogene enthält. Nicht verwendete DAB- Lösung
sollte entsprechend den Vorschriften entsorgt werden.
11. Die national verbindlichen Arbeitsvorschriften sind zu beachten.
12. Jeder schwerwiegende Zwischenfall, der im Zusammenhang mit-Histofine® DAB-2V aufgetreten ist, ist dem Hersteller und derzuständigen Behörde des Mitgliedstaats, in dem der Anwender und/oder der Patient niedergelassen ist, zu melden.

VII. REFERENZEN(1) Mokrỳ, et al. Versatility of immunohistochemical reactions: comprehensive survey of detection systems. Acta medica 1996 Apr
39:129-40(2) Yamada K, et al. In vitro assessment of antitumor immune responses using tumor antigen proteins produced by transgenic silkworms.
J Mater Sci Mater Med. 2021 May 17;32(6):58.

Katalog-/Codenummer

Temperaturbegrenzungen

Medizinische In Vitro
Diagnostik
Gerät

HERSTELLER

Batch-Code

Enthält ausreichend für䚾 N䚿 Tests

Zu verbrauchen bis

Anweisungen fürdie Verwendung konsultieren

Bevollmächtigter Vertreterin der Europäischen
Gemeinschaft

CE-Zeichen, Code derbenannte Stelle

Nur für IVD-Leistungsbewertung

MusterNICHIREI BIOSCIENCES INC.

6-19-20, Tsukiji, Chuo-ku, Tokyo 104-8402, JAPAN
Phone: 81-3-3248-2208, Facsimile: 81-3-3248-2243

MedEnvoy Global B.V.

Prinses Margrietplantsoen 33-Suite 1232595 AM The Hague
The Netherlands-Histofine ist ein geschütztes Warenzeichen von NICHIREI BIOSCIENCES. Die Registrierungsländer würden       TS-42531F-CE-G-v02