■External dimensions (Scale: mm)■Panel cutout (Scale: mm)

Mounting frame

44.5
□ 48

7.5

If horizontal close mounting is used for the controller,
IP66 speciﬁcation (Drip-proof/Dust-proof) may
be compromised, and all warranties will be
invalidated.

54.5
59.5（＊）■CT dimensions (Scale: mm)

3-phaseCTL-6-S-H (For 20A)

Power supply
100 to 240V AC or 24V AC/DC

l

15

0.5
7.5

2.8

Alarm unit＋−

K

CTL-6-S

21

10

40

30

Thermocouple
Electric
furnace

２

１

１３

２ 3A
３

EV1

250V
AC

NO

４ 3A
＋
O2/EV2

−

５

CT2

＋
O1

１４ −
１５
１６
１７

DI1 YB(＋)
COM SG

７

3A
250V
AC

15

■Console software SWS-ACS01M included

Using the Console software (SWS-ACS01M) with an USB communication
cable CMA (sold separately), parameters setting, logging and monitoring of
the controller can be performed by connecting to the USB port of the PC.

■USB communication cable CMA(Sold Separately) (Scale: mm)

USB plug

８

９

＋＋ A

１０

A
V

１１
B

１８ −−− B１２
DC TC RTD

Alarm 1 output

O2/EV2

Cooling output (D□ option), Alarm 2 output (A2 Option)
or Heater burnout alarm output (W, W3 option)

O1

Control output or Heating output (D□ option)

DC

Direct current or DC voltage input (For DC voltage input,+ side terminal number diﬀers depending on the voltage input.)TC

Thermocouple input

RTD

RTD input

CT1

CT input 1 (W, W3 option)

CT2

CT input 2 (W3 option)

DI

Contact input (SM option)

RS-485

Serial communication RS-485 (C5 option)！ ・This controller does not have a built-in power switch, circuit breaker and fuse.

It is necessary to install a power switch, circuit breaker and fuse near the controller.
・For a 24 V AC/DC power source, do not confuse polarity when using
direct current (DC).

50
2000

EV1

NO

＋

＞1V

DI2 YA(−)

RS250V DI 485
AC

NO

６

CT1

2-M3

35

！

100 to240V AC30

2-φ3.5200！

SAFETY
PRECAUTIONS●●To ensure safe and correct use, thoroughly read and understand the manual before using this instrument.
This instrument is intended to be used for industrial machinery, machine tools and measuring equipment. Verify
correct usage after purpose-of-use consultation with our agency or main oﬃce. (Never use this instrument for
medical purposes with which human lives are involved.)
External protection devices such as protective equipment against excessive temperature rise, etc. must be installed,
as malfunction of this product could result in serious damage to the system or injury to personnel. Proper periodic
maintenance is also required.
This instrument must be used under the conditions and environment described in the manual. Shinko Technos Co.,
Ltd. does not accept liability for any injury, loss of life or damage occurring due to the instrument being used underconditions not otherwise stated in the manual.

Caution with respect toExport Trade Control Ordinance
To avoid this instrument from being used as
a component in, or as being utilized in the
manufacture of weapons of mass destruction(i.e. military applications, military equipment,etc.), please investigate the end users andthe ﬁnal use of this instrument.
In the case of resale, ensure that this
instrument is not illegally exported.

SHINKO TECHNOS CO., LTD.
OVERSEAS DIVISIONUse a solderless terminal with an insulation sleeve in whichan M3 screw ﬁts. Tightening torque should be 0.63 N･m.
φ 3.2mm●●• This catalog is as of March 2018 and its contents are subject to change without notice.
• If you have any inquiries, please consult us or our agency.

■Solderless terminal

5.8 mm max.

−

１

φ12

* To prevent harmful eﬀects due to unexpected level
noise, it is recommended that a surge absorber be
installed between electromagnetic coils.

POWER SUPPLY

5.8 mm max.

＋

24V
AC/DC

40

20

Heater

L
40

Power supply
for alarm unit

25*

■Terminal arrangement
POWER
SUPPLYkk

Plug pin

POWER
SUPPLY

30

100

l
φ5.8

Electromagnetic
switchCTL-12-S36-10L1U (For 50A)

3

ACS-13A-R/M

Horizontal close mounting
n: Number of units mounted！ Caution:

45＋0.50(*) When a terminal cover (sold separately) is used

■Wiring example＋0.5

n×48−3 045＋0.5075

45＋0.50Terminal cover (*)

Gasket A

59.2

Alarm type, setting range and output type, etc. are the same as those of Alarm 1. See Alarm 1 output section.
If this option is ordered, Heating/Cooling control (D□ option) cannot be ordered.
This option and Heater burnout alarm (W, W3 option) utilize common output terminals.
Rating ------------------ Single-phase 20 A [W(20A)], Single-phase 50 A [W(50A)], 3-phase 20 A [W3(20A)], 3-phase 50 A [W3(50A)]
Must be speciﬁed.
Heater burnout alarm Setting range--------- Rated current 20 A [W(20A), W3(20A)]: 0.0 to 20.0 A
Rated current 50 A [W(50A), W3(50A)]: 0.0 to 50.0 A[W, W3 option]
Setting accuracy ---- Within ±5% of the rated value(CT included)
Action ------------------ ON/OFF action
Output ----------------- Relay contact 1a, Control capacity: 3A 250V AC (resistive load), Electrical life: 100,000 cycles
If this option is ordered, Heating/Cooling control (D□ option) cannot be ordered.
Heating control action: Same as the control output (OUT1)
Cooling control action
OUT2 proportional band (P) --- 0.0 to 10.0 times OUT1 proportional band (ON/OFF action when set to 0.0)
OUT2 integral time (I) ------------Same as that of OUT1
OUT2 derivative time (D) ------- Same as that of OUT1
OUT2 proportional cycle -------- 1 to 120 seconds
Heating/Cooling Overlap/Dead band --------------- Thermocouple, RTD: -100.0 to 100.0℃ (°F)
control
Direct current, voltage: -1000 to 1000 (The placement of the decimal point follows the selection.)[D□ option]
OUT2 hysteresis ------------------- 0.1 to 100.0℃ (°F), or 1 to 1000
OUT2 cooling method------------- One cooling method can be selected from Air cooling (linear characteristic), Oil cooling (1.5th power of thelinear characteristic) and Water cooling (2nd power of the linear characteristic) by keypad.
Output ------------------------------- Relay contact 1a, Control capacity: 3A 250V AC (resistive load), Electrical life: 100,000 cycles
Non-contact voltage 12 V DC±15% Max. 40 mA DC (short circuit protected)
If this option is ordered, Alarm 2 (A2 option) and Heater burnout alarm (W, W3 option) cannot be ordered.
Various operations (such as Reading and setting of various set values, Reading of PV and action status and Function change) can be
performed from an external PC.
Communication line ---------------- EIA RS-485
Communication method------------ Half-duplex communication
Synchronization method ---------- Start-stop synchronization
Communication speed ------------ 2400/4800/9600/19200 bps (Selectable by keypad)
Data bit /Parity ---------------------- Data bit: 7 or 8, Parity: Even/Odd /No parity (Selectable by keypad)
Stop bit --------------------------------- 1 or 2 (Selectable by keypad)
Communication protocol --------- Shinko protocol/Modbus ASCII/Modbus RTU (Selectable by keypad)
Number of connectable units --- Max. 31 units per host computer
Serial communication[C5 option]
Communication error detection -- Parity, checksum, LRC (Modbus ASCII), CRC-16 (Modbus RTU)
Data format
Communication protocol
Shinko protocol
Modbus ASCII
Modbus RTU
Start bit1
1
1
Data bit7
7 or 8
8
ParityYes (Even)
Yes (Even, Odd)
Yes (Even, Odd)
No parity
No parity
Stop bit1
1 or 2
1 or 2
If this option is ordered, Set value memory external selection (SM option) cannot be ordered.
SV1, SV2, SV3 or SV4 can be selected by the external contact.
The MEMO display indicates the selected memory number.
Set value
Contact input terminal DI2 can be used for ʻSet value memory external selectionʼ or for ʻOUT/OFF external selectionʼ in [Contact
memory external input function] in Setup mode.
selection
If ʻAuto/Manual control functionʼ is selected in [OUT/OFF key function] in Setup mode, externally Auto/Manual control can be switched.
[SM option]
Circuit current when closed: Approx. 12 mA
If this option is ordered, Serial communication (C5 option) cannot be ordered.
Alarm 2 output[A2 option]

47.5（＊）

When ordering, please specify an option code according to the userʼs needs.

10.5■Optional speciﬁcations

3.2mm

Reg. Oﬃce : 2-5-1, Senbahigashi, Minoo, Osaka, 562-0035, JapanTel: + 81 - 72 - 727 - 6100

Fax

: + 81 - 72 - 727 - 7006

URL

: http://www. shinko-technos. co. jp/e/E-mail: overseas @ shinko-technos. co. jp/e/201803 E06

1 Controller with the shortest depth (56 mm)

4 User friendly communication: Remote monitoring, data transmission

Point

Point

Higher functions and performance have been achieved with the shortest depth of 56 mm ACS-13A.
This provides cost and space reduction.

Standard Console communication function enables 1 to 1
communication between a PC with USB port and the ACS-13A.
Various settings and monitoring can be performed using theConsole software SWS-ACS01M. SWS-ACS01M is attached to
the USB communication cable CMA (sold separately).

Actual size

USB communication cable CMA

Displays of the Console software (SWS-ACS01M)■Serial Communication (RS-485) (C5 option)

Serial communication (RS-485) between a PC/Touch screen unit and the ACS-13A enables various settings and remote monitoring.
A maximum of 31 units in a centralized control system can be established.
Shinko protocol and Open Network Modbus protocol are usable.

●When monitoring multiple ACS-13A units with a
PC or Touch screen unit

The shortest depth

Please use a gasket to reinforce Drip-proof/Dust-proof function.
Depth of control panel interior when gasket is used: 54.5 mm

56

By connecting to the Touch screen unit, up to 31 points oftemperature control can be easily monitored.
For a PC with RS-232C, a communication converter is required.
Touch screen units corresponding to the ACS-13A are as shown below.
Digital Electronics Corp.: SP series, GP series, LT series
Hakko Electronics Co., Ltd.: V9 series, V8 series, TS series

2 An easier viewing display with status color indication

Point

The PV display color can be selected from
red, green and orange.
The PV display color can also change
continuously depending on deviation
between PV and SV, which allows easy
and distinct status checking.

●PV color continuous change mode

RS-232C

SIF-600

RS-485

ACS-13A

5 Standard Drip-proof/Dust-proof (IP66)●Alarm color (when setting High/Low limits alarm)

Low limit alarm active

Regular status

High limit alarm active

All segments lit

Point

・Can be used in a dust or water splashed environment.

This function is suitable for heaters which are not designed
for sudden changes in output. This is suited for controllingheaters such as the Kanthal Super. This also stabilizes
control by suppressing output ﬂuctuation.

■Name and functions of the sections
１１

３
４

SENS

A1LM

５
６

Versatile controls, speciﬁcations and enhancements■4-point SV, using external selection (SM option)
SV memory function, which can switch 4 points of
SV using external contact signal, is equipped.
After registering the SVs (from SV1 to SV4), they
can be easily switched by external operation.

■3-phase Heater burnout alarm (W3 option)

Any trouble in 3-phase heaters such as burnout or
deterioration can be detected by connecting 2 CTs(current transformers).
Heater burnout alarm types 20A and 50A are
available for both single phase and 3-phase.

6 Output Rate-of-Change (Standard)

Point

１

It is easier to see the SV and PV, using an 11-segment
LCD display.

PLCs corresponding to SIF-600 and its manufacturer:● Mitsubishi Electric Corp.
AJ71UC24, A1SJ71UC24-R2/R4/PRF,
A1SJ71C24-R2/R4/PRF, QJ71C24● Omron Corp.
CS1W-SCU21-V1, CJ1W-SCU21, CJ1W-SCU41● Keyence Corp.
KV-L20V● Yokogawa Electric Corp.
F3LC11-2N, F3LC11-1F, F3LC12-1F● Fuji Electric Co., Ltd.
NP1L-RS1, NP1L-RS2, NP1L-RS3, NP1L-RS4
For details about the above PLCs, please contact us or ournearest agency.

ACS-13A

２

3

PLC

Communication
converter
IF-400

PV is lower than [SV−PV color range]

７

１０

８

９

１２

ＡＣＳ-１３Ａ
Control
output(OUT1)
Input

□，□，
□，
□ ACS-13A（W48×H48×D62mm）
Relay contact 1a
Non-contact voltage (for SSR drive)
Direct current
Multi-input
100 to 240V AC＊
Power supply voltage
１
24V AC/DC
A2
Alarm 2
C5
Serial communication (EIA RS-485)
DR
Heating/Cooling control (relay output)
OUT2
DS
Heating/Cooling control (SSR output)SM
Set value memory external selection
OptionW (20A) Rated current: Single-phase 20A
HeaterW (50A) Rated current: Single-phase 50A
burnout
W3 (20A) Rated current: 3-phase 20A
Alarm
W3 (50A) Rated current: 3-phase 50A① PV indicator: Lights when the PV is indicated in PV/SV display mode.
② PV display: Indicates the PV (process variable).
③ SV indicator: Lights when the SV is indicated in PV/SV display mode.
④ MEMO indicator : Lights when Set value memory external selection (SM option) is ordered.
⑤ MEMO display : Indicates the Set value memory number.
⑥ SV display: Indicates the SV (desired value).
⑦ Increase key: Increases the numeric value.
⑧ Decrease key : Decreases the numeric value.
⑨ Mode key: Selects a setting mode, or registers the set value.
To register the set (selected) value, press this key.
⑩ OUT/OFF key : Switches control output ON/OFF or Auto/Manual control function.
⑪ Action indicators
O1(OUT1): Lights when control output is ON, or when Heating output (D□ option) is ON.
O2(OUT2): Lights when Cooling output (D□ option) is ON.
EV1 : Lights when Alarm 1 output is ON.
EV2 : Lights when Alarm 2 output (A2 option) is ON or when Heater burnout alarm (W, W3 option) is ON.
: Flashes while AT (auto-tuning) or auto-reset is performing.
AT
T/R : Lights when Serial communication (C5 option) is performing (TX output).
LOCK : Lights when Lock 1, Lock 2 or Lock 3 is selected.
⑫ Console connector: The following operations can be conducted by connecting to the USB
communication cable (Model: CMA, sold separately). (1) Reading and setting of SV, PID
and various set values, (2) Reading of PV and action status, (3) Function change

□／□
Ｒ
Ｓ
Ａ
Ｍ

Designate the speciﬁcation from the □ columns.
＊For the power supply voltage, 100 to 240V AC is standard, however, when ordering
24V AC/DC, enter "1" after the input code.
When ordering options, punctuate them with a comma.
・If A2 option is speciﬁed, DR or DS option cannot be ordered.
・If If C5 option is speciﬁed, SM option cannot be ordered.
・If D□ option is speciﬁed, A2, W or W3 option cannot be ordered.
・If W or W3 option is speciﬁed, DR or DS option cannot be ordered.
・For direct current output type, W or W3 option cannot be ordered.

■Rated range (Full multi-range)
Input type

Input range−200 to 1370 ℃ −320 to 2500 °F
Ｋ−200.0 to 400.0℃ −320.0 to 750.0 °FＪ−200 to 1000 ℃ −320 to 1800 °FＲ
0 to 1760 ℃
0 to 3200 °FＳ
0 to 1760 ℃
0 to 3200 °F
Thermocouple Ｂ
0 to 1820 ℃
0 to 3300 °F
Ｅ−200 to 800 ℃ −320 to 1500 °F
Ｔ−200.0 to 400.0℃ −320.0 to 750.0 °FＮ−200 to 1300 ℃ −320 to 2300 °F
PL-Ⅱ
0 to 1390 ℃
0 to 2500 °F
C(W/Re5-26)
0 to 2315 ℃
0 to 4200 °F−200 to 850 ℃ −320 to 1500 °F
Pt100−200.0 to 850.0℃ −320.0 to 1500.0°
F
RTD−200 to 500 ℃ −320 to 900°
F
JPt100−200.0 to 500.0℃ −320.0 to 900.0 °F
4 to 20mA DC
Direct current
0 to 20mA DC
0 to 1V DC−2000 to 10000
0 to 10V DC
DC voltage
1 to 5V DC
0 to 5V DC

・For Direct current and voltage input, scaling and decimal point place change
are possible.

■Standard speciﬁcations
PV display --------- 11-segment Backlight LCD Red/ Green /Orange, Character size, 12.0 x 5.4mm (H x W)
SV display --------- 11-segment Backlight LCD Green, Character size, 6.0 x 3.5mm (H x W)
MEMO display ---- 11-segment Backlight LCD Green, Character size, 4.8 x 2.8mm (H x W)
Indicators------------ Backlight Orange
Thermocouple ---- K, J, R, S, B, E, T, N, PL- Ⅱ, C (W/Re5-26) External resistance: 100Ω max., however, B input: 40Ω max.
RTD------------------- Pt100, JPt100: 3-wire type (Allowable input lead wire resistance: 10Ω max. per wire)
Direct current------ 0 to 20 mA DC, 4 to 20 mA DC: Input impedance: 50Ω , Allowable input current: 50 mA DC max.
Input
DC voltage--------- 0 to 1 V DC: Input impedance 1 MΩ min.
Allowable input voltage: 5 V DC max. Allowable signal source resistance: 2 kΩ max.
0 to 5 V DC, 1 to 5 V DC, 0 to 10 V DC: Input impedance: 100 kΩ min. Allowable input voltage: 15 V DC max.
Allowable signal source resistance: 100Ω max.
Thermocouple--------------- Within ±0.2% of each input span ±1digit, or within ±2℃ (4°F), whichever is greater
However, R, S input, the range is 0 to 200℃ (32 to 392°
F): Within ±6℃ (12°F)
Accuracy
B input, the range is 0 to 300℃ (32 to 572°
F): The accuracy is not guaranteed.
K, J, E, T, N input, less than 0℃ (32°
F): Within ±0.4% of input span ±1digit(Setting・Indication)
RTD----------------------------- Within ±0.1% of each input span±1digit, or within ±1℃ (2°F), whichever is greater
Direct current, DC voltage -- Within ±0.2% of each input span±1digit
Input sampling period 250 ms
Relay contact -----------1a, control capacity: 3 A 250 V AC (resistive load), 1 A 250 V AC (inductive load cosø=0.4), Electrical life: 100,000 cycles
Control output
Non-contact voltage -- 12 V DC ±15% Max. 40 mA DC (short circuit protected)(OUT1)
Direct current ----------- 4 to 20 mA DC Load resistance: Max. 550ΩThe following actions can be selected by keypad. (Factory default: PID)
PID (with auto-tuning function), PI, PD (with auto-reset function), P (with auto-reset function), ON/OFF
F), 0.0 to 1000.0℃ (1999.9°
F), or 0.0 to 100.0%(ON/OFF control when set to 0 or 0.0)
OUT1 proportional band (P) --------- 0 to 1000℃ (2000°
Integral time (I) --------------------------- 0 to 1000 seconds (OFF when set to 0)
Derivative time (D) ---------------------- 0 to 300 seconds (OFF when set to 0)
OUT1 ON/OFF action ----------------- When the proportional band is set to 0 or 0.0
Control action
OUT1 proportional cycle ------------- 1 to 120 seconds (Not available for Direct current output type)
ARW ----------------------------------------- 0 to 100%
OUT1 ON/OFF action hysteresis --- 0.1 to 100.0℃ (°F) or 1 to 1000
OUT1 high limit ------------------------- 0 to 100% (Direct current: -5 to 105%)
OUT1 low limit -------------------------- 0 to 100% (Direct current: -5 to 105%)
Alarm types can be selected by keypad. (Factory default: No alarm action)
・No alarm action
・High limit alarm(Deviation setting) Setting range: −(Input span) to Input span・Low limit alarm(Deviation setting) Setting range: −(Input span) to Input span・High/Low limits alarm(Deviation setting) Setting range: 0 to Input span・High/Low limit range alarm(Deviation setting) Setting range: 0 to Input span・Process high alarm
Setting range: Input range low limit value to Input range high limit value
・Process low alarm
Setting range: Input range low limit value to Input range high limit value
Alarm 1 output
・High limit with standby alarm(Deviation setting) Setting range: −(Input span) to Input span・Low limit with standby alarm(Deviation setting) Setting range: −(Input span) to Input span・High/Low limits with standby alarm (Deviation setting) Setting range: 0 to Input span
Setting accuracy----- Same as indication accuracy
Action-------------------- ON/OFF action
Hysteresis-------------- 0.1 to 100.0℃ (°F) or 1 to 1000
Output------------------- Relay contact 1a, Control capacity: 3 A 250 V AC (Resistive load), Electrical life: 100,000 cycles
100 to 240 V AC 50/60 Hz (Allowable voltage ﬂuctuation range: 85 to 264 V AC)
Supply voltage
24 V AC/DC 50/60 Hz (Allowable voltage ﬂuctuation range: 20 to 28 V AC/DC)Power consumption Approx. 8VA
Insulation resistance 10 MΩ min., at 500 V DC
Dielectric strength Between Input terminal and Power terminal: 1.5 kV AC for 1 minute Between Output terminal and Power terminal: 1.5 kV AC for 1 minuteEnvironment
Ambient temperature: 0 to 50℃ Ambient humidity: 35 to 85 %RH (non-condensing),Drip-proof/Dust-proof IP66 (for front panel only)
Safety standards UL: Power input rating 100-240 V AC, 24 V AC/DC File No. E159038
Case (Material, Color) Material: Flame resistant resin Color: Black
Mounting method Mounting frame (Mountable control panel thickness: 1 to 5 mm)
Setting method Sheet key input
External dimensions W48 x H48 x D62 mm (Depth of control panel interior when the gasket is used: 54.5 mm)
WeightApprox. 120g
Power failure countermeasure, Self-diagnosis, Automatic cold junction temperature compensation (only for thermocouple), Burnout,
Attached functions
Input error indication, Indication range, Control range, Warm-up indication, Auto/Manual control switching, Console communication
Mounting frame 1 piece, Gasket A (Front mounted to the ACS-13A) 1 piece
Instruction manual (A3 unfolded, English/Japanese) 1 copy
Accessories included CT (Current transformer): CTL-6-S-H: 1 piece [W (20A) option],
CTL-12-S36-10L1U: 1 piece [W (50A) option]
CTL-6-S-H: 2 pieces [W3 (20A) option], CTL-12-S36-10L1U: 2 pieces [W3 (50A) option]
Accessories sold separately Terminal cover (TC-ACS), USB communication cable CMA (to which the Console software SWS-ACS01M is attached)
Display

PV is higher than [SV＋PV color range]

RS-485

Point

By connecting to a PLC via PLC interface unit SIF-600,
up to 32 units of the ACS-13A can be connected.

Touch screen unitPC

PV is within [SV±PV color range]

7 Types of PV Display Color
・Regular status: Green・Regular status: Red
・Regular status: Orange
・Alarm OFF: Green, Alarm ON: Red・Alarm OFF: Orange, Alarm ON: Red・PV color changes continuously :OrangeGreenRed・PV color changes continuously + Alarm ON (Red)●When communicating with a PLC and up to 32
units of the ACS-13A■Model

1 Controller with the shortest depth (56 mm)

4 User friendly communication: Remote monitoring, data transmission

Point

Point

Higher functions and performance have been achieved with the shortest depth of 56 mm ACS-13A.
This provides cost and space reduction.

Standard Console communication function enables 1 to 1
communication between a PC with USB port and the ACS-13A.
Various settings and monitoring can be performed using theConsole software SWS-ACS01M. SWS-ACS01M is attached to
the USB communication cable CMA (sold separately).

Actual size

USB communication cable CMA

Displays of the Console software (SWS-ACS01M)■Serial Communication (RS-485) (C5 option)

Serial communication (RS-485) between a PC/Touch screen unit and the ACS-13A enables various settings and remote monitoring.
A maximum of 31 units in a centralized control system can be established.
Shinko protocol and Open Network Modbus protocol are usable.

●When monitoring multiple ACS-13A units with a
PC or Touch screen unit

The shortest depth

Please use a gasket to reinforce Drip-proof/Dust-proof function.
Depth of control panel interior when gasket is used: 54.5 mm

56

By connecting to the Touch screen unit, up to 31 points oftemperature control can be easily monitored.
For a PC with RS-232C, a communication converter is required.
Touch screen units corresponding to the ACS-13A are as shown below.
Digital Electronics Corp.: SP series, GP series, LT series
Hakko Electronics Co., Ltd.: V9 series, V8 series, TS series

2 An easier viewing display with status color indication

Point

The PV display color can be selected from
red, green and orange.
The PV display color can also change
continuously depending on deviation
between PV and SV, which allows easy
and distinct status checking.

●PV color continuous change mode

RS-232C

SIF-600

RS-485

ACS-13A

5 Standard Drip-proof/Dust-proof (IP66)●Alarm color (when setting High/Low limits alarm)

Low limit alarm active

Regular status

High limit alarm active

All segments lit

Point

・Can be used in a dust or water splashed environment.

This function is suitable for heaters which are not designed
for sudden changes in output. This is suited for controllingheaters such as the Kanthal Super. This also stabilizes
control by suppressing output ﬂuctuation.

■Name and functions of the sections
１１

３
４

SENS

A1LM

５
６

Versatile controls, speciﬁcations and enhancements■4-point SV, using external selection (SM option)
SV memory function, which can switch 4 points of
SV using external contact signal, is equipped.
After registering the SVs (from SV1 to SV4), they
can be easily switched by external operation.

■3-phase Heater burnout alarm (W3 option)

Any trouble in 3-phase heaters such as burnout or
deterioration can be detected by connecting 2 CTs(current transformers).
Heater burnout alarm types 20A and 50A are
available for both single phase and 3-phase.

6 Output Rate-of-Change (Standard)

Point

１

It is easier to see the SV and PV, using an 11-segment
LCD display.

PLCs corresponding to SIF-600 and its manufacturer:● Mitsubishi Electric Corp.
AJ71UC24, A1SJ71UC24-R2/R4/PRF,
A1SJ71C24-R2/R4/PRF, QJ71C24● Omron Corp.
CS1W-SCU21-V1, CJ1W-SCU21, CJ1W-SCU41● Keyence Corp.
KV-L20V● Yokogawa Electric Corp.
F3LC11-2N, F3LC11-1F, F3LC12-1F● Fuji Electric Co., Ltd.
NP1L-RS1, NP1L-RS2, NP1L-RS3, NP1L-RS4
For details about the above PLCs, please contact us or ournearest agency.

ACS-13A

２

3

PLC

Communication
converter
IF-400

PV is lower than [SV−PV color range]

７

１０

８

９

１２

ＡＣＳ-１３Ａ
Control
output(OUT1)
Input

□，□，
□，
□ ACS-13A（W48×H48×D62mm）
Relay contact 1a
Non-contact voltage (for SSR drive)
Direct current
Multi-input
100 to 240V AC＊
Power supply voltage
１
24V AC/DC
A2
Alarm 2
C5
Serial communication (EIA RS-485)
DR
Heating/Cooling control (relay output)
OUT2
DS
Heating/Cooling control (SSR output)SM
Set value memory external selection
OptionW (20A) Rated current: Single-phase 20A
HeaterW (50A) Rated current: Single-phase 50A
burnout
W3 (20A) Rated current: 3-phase 20A
Alarm
W3 (50A) Rated current: 3-phase 50A① PV indicator: Lights when the PV is indicated in PV/SV display mode.
② PV display: Indicates the PV (process variable).
③ SV indicator: Lights when the SV is indicated in PV/SV display mode.
④ MEMO indicator : Lights when Set value memory external selection (SM option) is ordered.
⑤ MEMO display : Indicates the Set value memory number.
⑥ SV display: Indicates the SV (desired value).
⑦ Increase key: Increases the numeric value.
⑧ Decrease key : Decreases the numeric value.
⑨ Mode key: Selects a setting mode, or registers the set value.
To register the set (selected) value, press this key.
⑩ OUT/OFF key : Switches control output ON/OFF or Auto/Manual control function.
⑪ Action indicators
O1(OUT1): Lights when control output is ON, or when Heating output (D□ option) is ON.
O2(OUT2): Lights when Cooling output (D□ option) is ON.
EV1 : Lights when Alarm 1 output is ON.
EV2 : Lights when Alarm 2 output (A2 option) is ON or when Heater burnout alarm (W, W3 option) is ON.
: Flashes while AT (auto-tuning) or auto-reset is performing.
AT
T/R : Lights when Serial communication (C5 option) is performing (TX output).
LOCK : Lights when Lock 1, Lock 2 or Lock 3 is selected.
⑫ Console connector: The following operations can be conducted by connecting to the USB
communication cable (Model: CMA, sold separately). (1) Reading and setting of SV, PID
and various set values, (2) Reading of PV and action status, (3) Function change

□／□
Ｒ
Ｓ
Ａ
Ｍ

Designate the speciﬁcation from the □ columns.
＊For the power supply voltage, 100 to 240V AC is standard, however, when ordering
24V AC/DC, enter "1" after the input code.
When ordering options, punctuate them with a comma.
・If A2 option is speciﬁed, DR or DS option cannot be ordered.
・If If C5 option is speciﬁed, SM option cannot be ordered.
・If D□ option is speciﬁed, A2, W or W3 option cannot be ordered.
・If W or W3 option is speciﬁed, DR or DS option cannot be ordered.
・For direct current output type, W or W3 option cannot be ordered.

■Rated range (Full multi-range)
Input type

Input range−200 to 1370 ℃ −320 to 2500 °F
Ｋ−200.0 to 400.0℃ −320.0 to 750.0 °FＪ−200 to 1000 ℃ −320 to 1800 °FＲ
0 to 1760 ℃
0 to 3200 °FＳ
0 to 1760 ℃
0 to 3200 °F
Thermocouple Ｂ
0 to 1820 ℃
0 to 3300 °F
Ｅ−200 to 800 ℃ −320 to 1500 °F
Ｔ−200.0 to 400.0℃ −320.0 to 750.0 °FＮ−200 to 1300 ℃ −320 to 2300 °F
PL-Ⅱ
0 to 1390 ℃
0 to 2500 °F
C(W/Re5-26)
0 to 2315 ℃
0 to 4200 °F−200 to 850 ℃ −320 to 1500 °F
Pt100−200.0 to 850.0℃ −320.0 to 1500.0°
F
RTD−200 to 500 ℃ −320 to 900°
F
JPt100−200.0 to 500.0℃ −320.0 to 900.0 °F
4 to 20mA DC
Direct current
0 to 20mA DC
0 to 1V DC−2000 to 10000
0 to 10V DC
DC voltage
1 to 5V DC
0 to 5V DC

・For Direct current and voltage input, scaling and decimal point place change
are possible.

■Standard speciﬁcations
PV display --------- 11-segment Backlight LCD Red/ Green /Orange, Character size, 12.0 x 5.4mm (H x W)
SV display --------- 11-segment Backlight LCD Green, Character size, 6.0 x 3.5mm (H x W)
MEMO display ---- 11-segment Backlight LCD Green, Character size, 4.8 x 2.8mm (H x W)
Indicators------------ Backlight Orange
Thermocouple ---- K, J, R, S, B, E, T, N, PL- Ⅱ, C (W/Re5-26) External resistance: 100Ω max., however, B input: 40Ω max.
RTD------------------- Pt100, JPt100: 3-wire type (Allowable input lead wire resistance: 10Ω max. per wire)
Direct current------ 0 to 20 mA DC, 4 to 20 mA DC: Input impedance: 50Ω , Allowable input current: 50 mA DC max.
Input
DC voltage--------- 0 to 1 V DC: Input impedance 1 MΩ min.
Allowable input voltage: 5 V DC max. Allowable signal source resistance: 2 kΩ max.
0 to 5 V DC, 1 to 5 V DC, 0 to 10 V DC: Input impedance: 100 kΩ min. Allowable input voltage: 15 V DC max.
Allowable signal source resistance: 100Ω max.
Thermocouple--------------- Within ±0.2% of each input span ±1digit, or within ±2℃ (4°F), whichever is greater
However, R, S input, the range is 0 to 200℃ (32 to 392°
F): Within ±6℃ (12°F)
Accuracy
B input, the range is 0 to 300℃ (32 to 572°
F): The accuracy is not guaranteed.
K, J, E, T, N input, less than 0℃ (32°
F): Within ±0.4% of input span ±1digit(Setting・Indication)
RTD----------------------------- Within ±0.1% of each input span±1digit, or within ±1℃ (2°F), whichever is greater
Direct current, DC voltage -- Within ±0.2% of each input span±1digit
Input sampling period 250 ms
Relay contact -----------1a, control capacity: 3 A 250 V AC (resistive load), 1 A 250 V AC (inductive load cosø=0.4), Electrical life: 100,000 cycles
Control output
Non-contact voltage -- 12 V DC ±15% Max. 40 mA DC (short circuit protected)(OUT1)
Direct current ----------- 4 to 20 mA DC Load resistance: Max. 550ΩThe following actions can be selected by keypad. (Factory default: PID)
PID (with auto-tuning function), PI, PD (with auto-reset function), P (with auto-reset function), ON/OFF
F), 0.0 to 1000.0℃ (1999.9°
F), or 0.0 to 100.0%(ON/OFF control when set to 0 or 0.0)
OUT1 proportional band (P) --------- 0 to 1000℃ (2000°
Integral time (I) --------------------------- 0 to 1000 seconds (OFF when set to 0)
Derivative time (D) ---------------------- 0 to 300 seconds (OFF when set to 0)
OUT1 ON/OFF action ----------------- When the proportional band is set to 0 or 0.0
Control action
OUT1 proportional cycle ------------- 1 to 120 seconds (Not available for Direct current output type)
ARW ----------------------------------------- 0 to 100%
OUT1 ON/OFF action hysteresis --- 0.1 to 100.0℃ (°F) or 1 to 1000
OUT1 high limit ------------------------- 0 to 100% (Direct current: -5 to 105%)
OUT1 low limit -------------------------- 0 to 100% (Direct current: -5 to 105%)
Alarm types can be selected by keypad. (Factory default: No alarm action)
・No alarm action
・High limit alarm(Deviation setting) Setting range: −(Input span) to Input span・Low limit alarm(Deviation setting) Setting range: −(Input span) to Input span・High/Low limits alarm(Deviation setting) Setting range: 0 to Input span・High/Low limit range alarm(Deviation setting) Setting range: 0 to Input span・Process high alarm
Setting range: Input range low limit value to Input range high limit value
・Process low alarm
Setting range: Input range low limit value to Input range high limit value
Alarm 1 output
・High limit with standby alarm(Deviation setting) Setting range: −(Input span) to Input span・Low limit with standby alarm(Deviation setting) Setting range: −(Input span) to Input span・High/Low limits with standby alarm (Deviation setting) Setting range: 0 to Input span
Setting accuracy----- Same as indication accuracy
Action-------------------- ON/OFF action
Hysteresis-------------- 0.1 to 100.0℃ (°F) or 1 to 1000
Output------------------- Relay contact 1a, Control capacity: 3 A 250 V AC (Resistive load), Electrical life: 100,000 cycles
100 to 240 V AC 50/60 Hz (Allowable voltage ﬂuctuation range: 85 to 264 V AC)
Supply voltage
24 V AC/DC 50/60 Hz (Allowable voltage ﬂuctuation range: 20 to 28 V AC/DC)Power consumption Approx. 8VA
Insulation resistance 10 MΩ min., at 500 V DC
Dielectric strength Between Input terminal and Power terminal: 1.5 kV AC for 1 minute Between Output terminal and Power terminal: 1.5 kV AC for 1 minuteEnvironment
Ambient temperature: 0 to 50℃ Ambient humidity: 35 to 85 %RH (non-condensing),Drip-proof/Dust-proof IP66 (for front panel only)
Safety standards UL: Power input rating 100-240 V AC, 24 V AC/DC File No. E159038
Case (Material, Color) Material: Flame resistant resin Color: Black
Mounting method Mounting frame (Mountable control panel thickness: 1 to 5 mm)
Setting method Sheet key input
External dimensions W48 x H48 x D62 mm (Depth of control panel interior when the gasket is used: 54.5 mm)
WeightApprox. 120g
Power failure countermeasure, Self-diagnosis, Automatic cold junction temperature compensation (only for thermocouple), Burnout,
Attached functions
Input error indication, Indication range, Control range, Warm-up indication, Auto/Manual control switching, Console communication
Mounting frame 1 piece, Gasket A (Front mounted to the ACS-13A) 1 piece
Instruction manual (A3 unfolded, English/Japanese) 1 copy
Accessories included CT (Current transformer): CTL-6-S-H: 1 piece [W (20A) option],
CTL-12-S36-10L1U: 1 piece [W (50A) option]
CTL-6-S-H: 2 pieces [W3 (20A) option], CTL-12-S36-10L1U: 2 pieces [W3 (50A) option]
Accessories sold separately Terminal cover (TC-ACS), USB communication cable CMA (to which the Console software SWS-ACS01M is attached)
Display

PV is higher than [SV＋PV color range]

RS-485

Point

By connecting to a PLC via PLC interface unit SIF-600,
up to 32 units of the ACS-13A can be connected.

Touch screen unitPC

PV is within [SV±PV color range]

7 Types of PV Display Color
・Regular status: Green・Regular status: Red
・Regular status: Orange
・Alarm OFF: Green, Alarm ON: Red・Alarm OFF: Orange, Alarm ON: Red・PV color changes continuously :OrangeGreenRed・PV color changes continuously + Alarm ON (Red)●When communicating with a PLC and up to 32
units of the ACS-13A■Model

1 Controller with the shortest depth (56 mm)

4 User friendly communication: Remote monitoring, data transmission

Point

Point

Higher functions and performance have been achieved with the shortest depth of 56 mm ACS-13A.
This provides cost and space reduction.

Standard Console communication function enables 1 to 1
communication between a PC with USB port and the ACS-13A.
Various settings and monitoring can be performed using theConsole software SWS-ACS01M. SWS-ACS01M is attached to
the USB communication cable CMA (sold separately).

Actual size

USB communication cable CMA

Displays of the Console software (SWS-ACS01M)■Serial Communication (RS-485) (C5 option)

Serial communication (RS-485) between a PC/Touch screen unit and the ACS-13A enables various settings and remote monitoring.
A maximum of 31 units in a centralized control system can be established.
Shinko protocol and Open Network Modbus protocol are usable.

●When monitoring multiple ACS-13A units with a
PC or Touch screen unit

The shortest depth

Please use a gasket to reinforce Drip-proof/Dust-proof function.
Depth of control panel interior when gasket is used: 54.5 mm

56

By connecting to the Touch screen unit, up to 31 points oftemperature control can be easily monitored.
For a PC with RS-232C, a communication converter is required.
Touch screen units corresponding to the ACS-13A are as shown below.
Digital Electronics Corp.: SP series, GP series, LT series
Hakko Electronics Co., Ltd.: V9 series, V8 series, TS series

2 An easier viewing display with status color indication

Point

The PV display color can be selected from
red, green and orange.
The PV display color can also change
continuously depending on deviation
between PV and SV, which allows easy
and distinct status checking.

●PV color continuous change mode

RS-232C

SIF-600

RS-485

ACS-13A

5 Standard Drip-proof/Dust-proof (IP66)●Alarm color (when setting High/Low limits alarm)

Low limit alarm active

Regular status

High limit alarm active

All segments lit

Point

・Can be used in a dust or water splashed environment.

This function is suitable for heaters which are not designed
for sudden changes in output. This is suited for controllingheaters such as the Kanthal Super. This also stabilizes
control by suppressing output ﬂuctuation.

■Name and functions of the sections
１１

３
４

SENS

A1LM

５
６

Versatile controls, speciﬁcations and enhancements■4-point SV, using external selection (SM option)
SV memory function, which can switch 4 points of
SV using external contact signal, is equipped.
After registering the SVs (from SV1 to SV4), they
can be easily switched by external operation.

■3-phase Heater burnout alarm (W3 option)

Any trouble in 3-phase heaters such as burnout or
deterioration can be detected by connecting 2 CTs(current transformers).
Heater burnout alarm types 20A and 50A are
available for both single phase and 3-phase.

6 Output Rate-of-Change (Standard)

Point

１

It is easier to see the SV and PV, using an 11-segment
LCD display.

PLCs corresponding to SIF-600 and its manufacturer:● Mitsubishi Electric Corp.
AJ71UC24, A1SJ71UC24-R2/R4/PRF,
A1SJ71C24-R2/R4/PRF, QJ71C24● Omron Corp.
CS1W-SCU21-V1, CJ1W-SCU21, CJ1W-SCU41● Keyence Corp.
KV-L20V● Yokogawa Electric Corp.
F3LC11-2N, F3LC11-1F, F3LC12-1F● Fuji Electric Co., Ltd.
NP1L-RS1, NP1L-RS2, NP1L-RS3, NP1L-RS4
For details about the above PLCs, please contact us or ournearest agency.

ACS-13A

２

3

PLC

Communication
converter
IF-400

PV is lower than [SV−PV color range]

７

１０

８

９

１２

ＡＣＳ-１３Ａ
Control
output(OUT1)
Input

□，□，
□，
□ ACS-13A（W48×H48×D62mm）
Relay contact 1a
Non-contact voltage (for SSR drive)
Direct current
Multi-input
100 to 240V AC＊
Power supply voltage
１
24V AC/DC
A2
Alarm 2
C5
Serial communication (EIA RS-485)
DR
Heating/Cooling control (relay output)
OUT2
DS
Heating/Cooling control (SSR output)SM
Set value memory external selection
OptionW (20A) Rated current: Single-phase 20A
HeaterW (50A) Rated current: Single-phase 50A
burnout
W3 (20A) Rated current: 3-phase 20A
Alarm
W3 (50A) Rated current: 3-phase 50A① PV indicator: Lights when the PV is indicated in PV/SV display mode.
② PV display: Indicates the PV (process variable).
③ SV indicator: Lights when the SV is indicated in PV/SV display mode.
④ MEMO indicator : Lights when Set value memory external selection (SM option) is ordered.
⑤ MEMO display : Indicates the Set value memory number.
⑥ SV display: Indicates the SV (desired value).
⑦ Increase key: Increases the numeric value.
⑧ Decrease key : Decreases the numeric value.
⑨ Mode key: Selects a setting mode, or registers the set value.
To register the set (selected) value, press this key.
⑩ OUT/OFF key : Switches control output ON/OFF or Auto/Manual control function.
⑪ Action indicators
O1(OUT1): Lights when control output is ON, or when Heating output (D□ option) is ON.
O2(OUT2): Lights when Cooling output (D□ option) is ON.
EV1 : Lights when Alarm 1 output is ON.
EV2 : Lights when Alarm 2 output (A2 option) is ON or when Heater burnout alarm (W, W3 option) is ON.
: Flashes while AT (auto-tuning) or auto-reset is performing.
AT
T/R : Lights when Serial communication (C5 option) is performing (TX output).
LOCK : Lights when Lock 1, Lock 2 or Lock 3 is selected.
⑫ Console connector: The following operations can be conducted by connecting to the USB
communication cable (Model: CMA, sold separately). (1) Reading and setting of SV, PID
and various set values, (2) Reading of PV and action status, (3) Function change

□／□
Ｒ
Ｓ
Ａ
Ｍ

Designate the speciﬁcation from the □ columns.
＊For the power supply voltage, 100 to 240V AC is standard, however, when ordering
24V AC/DC, enter "1" after the input code.
When ordering options, punctuate them with a comma.
・If A2 option is speciﬁed, DR or DS option cannot be ordered.
・If If C5 option is speciﬁed, SM option cannot be ordered.
・If D□ option is speciﬁed, A2, W or W3 option cannot be ordered.
・If W or W3 option is speciﬁed, DR or DS option cannot be ordered.
・For direct current output type, W or W3 option cannot be ordered.

■Rated range (Full multi-range)
Input type

Input range−200 to 1370 ℃ −320 to 2500 °F
Ｋ−200.0 to 400.0℃ −320.0 to 750.0 °FＪ−200 to 1000 ℃ −320 to 1800 °FＲ
0 to 1760 ℃
0 to 3200 °FＳ
0 to 1760 ℃
0 to 3200 °F
Thermocouple Ｂ
0 to 1820 ℃
0 to 3300 °F
Ｅ−200 to 800 ℃ −320 to 1500 °F
Ｔ−200.0 to 400.0℃ −320.0 to 750.0 °FＮ−200 to 1300 ℃ −320 to 2300 °F
PL-Ⅱ
0 to 1390 ℃
0 to 2500 °F
C(W/Re5-26)
0 to 2315 ℃
0 to 4200 °F−200 to 850 ℃ −320 to 1500 °F
Pt100−200.0 to 850.0℃ −320.0 to 1500.0°
F
RTD−200 to 500 ℃ −320 to 900°
F
JPt100−200.0 to 500.0℃ −320.0 to 900.0 °F
4 to 20mA DC
Direct current
0 to 20mA DC
0 to 1V DC−2000 to 10000
0 to 10V DC
DC voltage
1 to 5V DC
0 to 5V DC

・For Direct current and voltage input, scaling and decimal point place change
are possible.

■Standard speciﬁcations
PV display --------- 11-segment Backlight LCD Red/ Green /Orange, Character size, 12.0 x 5.4mm (H x W)
SV display --------- 11-segment Backlight LCD Green, Character size, 6.0 x 3.5mm (H x W)
MEMO display ---- 11-segment Backlight LCD Green, Character size, 4.8 x 2.8mm (H x W)
Indicators------------ Backlight Orange
Thermocouple ---- K, J, R, S, B, E, T, N, PL- Ⅱ, C (W/Re5-26) External resistance: 100Ω max., however, B input: 40Ω max.
RTD------------------- Pt100, JPt100: 3-wire type (Allowable input lead wire resistance: 10Ω max. per wire)
Direct current------ 0 to 20 mA DC, 4 to 20 mA DC: Input impedance: 50Ω , Allowable input current: 50 mA DC max.
Input
DC voltage--------- 0 to 1 V DC: Input impedance 1 MΩ min.
Allowable input voltage: 5 V DC max. Allowable signal source resistance: 2 kΩ max.
0 to 5 V DC, 1 to 5 V DC, 0 to 10 V DC: Input impedance: 100 kΩ min. Allowable input voltage: 15 V DC max.
Allowable signal source resistance: 100Ω max.
Thermocouple--------------- Within ±0.2% of each input span ±1digit, or within ±2℃ (4°F), whichever is greater
However, R, S input, the range is 0 to 200℃ (32 to 392°
F): Within ±6℃ (12°F)
Accuracy
B input, the range is 0 to 300℃ (32 to 572°
F): The accuracy is not guaranteed.
K, J, E, T, N input, less than 0℃ (32°
F): Within ±0.4% of input span ±1digit(Setting・Indication)
RTD----------------------------- Within ±0.1% of each input span±1digit, or within ±1℃ (2°F), whichever is greater
Direct current, DC voltage -- Within ±0.2% of each input span±1digit
Input sampling period 250 ms
Relay contact -----------1a, control capacity: 3 A 250 V AC (resistive load), 1 A 250 V AC (inductive load cosø=0.4), Electrical life: 100,000 cycles
Control output
Non-contact voltage -- 12 V DC ±15% Max. 40 mA DC (short circuit protected)(OUT1)
Direct current ----------- 4 to 20 mA DC Load resistance: Max. 550ΩThe following actions can be selected by keypad. (Factory default: PID)
PID (with auto-tuning function), PI, PD (with auto-reset function), P (with auto-reset function), ON/OFF
F), 0.0 to 1000.0℃ (1999.9°
F), or 0.0 to 100.0%(ON/OFF control when set to 0 or 0.0)
OUT1 proportional band (P) --------- 0 to 1000℃ (2000°
Integral time (I) --------------------------- 0 to 1000 seconds (OFF when set to 0)
Derivative time (D) ---------------------- 0 to 300 seconds (OFF when set to 0)
OUT1 ON/OFF action ----------------- When the proportional band is set to 0 or 0.0
Control action
OUT1 proportional cycle ------------- 1 to 120 seconds (Not available for Direct current output type)
ARW ----------------------------------------- 0 to 100%
OUT1 ON/OFF action hysteresis --- 0.1 to 100.0℃ (°F) or 1 to 1000
OUT1 high limit ------------------------- 0 to 100% (Direct current: -5 to 105%)
OUT1 low limit -------------------------- 0 to 100% (Direct current: -5 to 105%)
Alarm types can be selected by keypad. (Factory default: No alarm action)
・No alarm action
・High limit alarm(Deviation setting) Setting range: −(Input span) to Input span・Low limit alarm(Deviation setting) Setting range: −(Input span) to Input span・High/Low limits alarm(Deviation setting) Setting range: 0 to Input span・High/Low limit range alarm(Deviation setting) Setting range: 0 to Input span・Process high alarm
Setting range: Input range low limit value to Input range high limit value
・Process low alarm
Setting range: Input range low limit value to Input range high limit value
Alarm 1 output
・High limit with standby alarm(Deviation setting) Setting range: −(Input span) to Input span・Low limit with standby alarm(Deviation setting) Setting range: −(Input span) to Input span・High/Low limits with standby alarm (Deviation setting) Setting range: 0 to Input span
Setting accuracy----- Same as indication accuracy
Action-------------------- ON/OFF action
Hysteresis-------------- 0.1 to 100.0℃ (°F) or 1 to 1000
Output------------------- Relay contact 1a, Control capacity: 3 A 250 V AC (Resistive load), Electrical life: 100,000 cycles
100 to 240 V AC 50/60 Hz (Allowable voltage ﬂuctuation range: 85 to 264 V AC)
Supply voltage
24 V AC/DC 50/60 Hz (Allowable voltage ﬂuctuation range: 20 to 28 V AC/DC)Power consumption Approx. 8VA
Insulation resistance 10 MΩ min., at 500 V DC
Dielectric strength Between Input terminal and Power terminal: 1.5 kV AC for 1 minute Between Output terminal and Power terminal: 1.5 kV AC for 1 minuteEnvironment
Ambient temperature: 0 to 50℃ Ambient humidity: 35 to 85 %RH (non-condensing),Drip-proof/Dust-proof IP66 (for front panel only)
Safety standards UL: Power input rating 100-240 V AC, 24 V AC/DC File No. E159038
Case (Material, Color) Material: Flame resistant resin Color: Black
Mounting method Mounting frame (Mountable control panel thickness: 1 to 5 mm)
Setting method Sheet key input
External dimensions W48 x H48 x D62 mm (Depth of control panel interior when the gasket is used: 54.5 mm)
WeightApprox. 120g
Power failure countermeasure, Self-diagnosis, Automatic cold junction temperature compensation (only for thermocouple), Burnout,
Attached functions
Input error indication, Indication range, Control range, Warm-up indication, Auto/Manual control switching, Console communication
Mounting frame 1 piece, Gasket A (Front mounted to the ACS-13A) 1 piece
Instruction manual (A3 unfolded, English/Japanese) 1 copy
Accessories included CT (Current transformer): CTL-6-S-H: 1 piece [W (20A) option],
CTL-12-S36-10L1U: 1 piece [W (50A) option]
CTL-6-S-H: 2 pieces [W3 (20A) option], CTL-12-S36-10L1U: 2 pieces [W3 (50A) option]
Accessories sold separately Terminal cover (TC-ACS), USB communication cable CMA (to which the Console software SWS-ACS01M is attached)
Display

PV is higher than [SV＋PV color range]

RS-485

Point

By connecting to a PLC via PLC interface unit SIF-600,
up to 32 units of the ACS-13A can be connected.

Touch screen unitPC

PV is within [SV±PV color range]

7 Types of PV Display Color
・Regular status: Green・Regular status: Red
・Regular status: Orange
・Alarm OFF: Green, Alarm ON: Red・Alarm OFF: Orange, Alarm ON: Red・PV color changes continuously :OrangeGreenRed・PV color changes continuously + Alarm ON (Red)●When communicating with a PLC and up to 32
units of the ACS-13A■Model■External dimensions (Scale: mm)■Panel cutout (Scale: mm)

Mounting frame

44.5
□ 48

7.5

If horizontal close mounting is used for the controller,
IP66 speciﬁcation (Drip-proof/Dust-proof) may
be compromised, and all warranties will be
invalidated.

54.5
59.5（＊）■CT dimensions (Scale: mm)

3-phaseCTL-6-S-H (For 20A)

Power supply
100 to 240V AC or 24V AC/DC

l

15

0.5
7.5

2.8

Alarm unit＋−

K

CTL-6-S

21

10

40

30

Thermocouple
Electric
furnace

２

１

１３

２ 3A
３

EV1

250V
AC

NO

４ 3A
＋
O2/EV2

−

５

CT2

＋
O1

１４ −
１５
１６
１７

DI1 YB(＋)
COM SG

７

3A
250V
AC

15

■Console software SWS-ACS01M included

Using the Console software (SWS-ACS01M) with an USB communication
cable CMA (sold separately), parameters setting, logging and monitoring of
the controller can be performed by connecting to the USB port of the PC.

■USB communication cable CMA(Sold Separately) (Scale: mm)

USB plug

８

９

＋＋ A

１０

A
V

１１
B

１８ −−− B１２
DC TC RTD

Alarm 1 output

O2/EV2

Cooling output (D□ option), Alarm 2 output (A2 Option)
or Heater burnout alarm output (W, W3 option)

O1

Control output or Heating output (D□ option)

DC

Direct current or DC voltage input (For DC voltage input,+ side terminal number diﬀers depending on the voltage input.)TC

Thermocouple input

RTD

RTD input

CT1

CT input 1 (W, W3 option)

CT2

CT input 2 (W3 option)

DI

Contact input (SM option)

RS-485

Serial communication RS-485 (C5 option)！ ・This controller does not have a built-in power switch, circuit breaker and fuse.

It is necessary to install a power switch, circuit breaker and fuse near the controller.
・For a 24 V AC/DC power source, do not confuse polarity when using
direct current (DC).

50
2000

EV1

NO

＋

＞1V

DI2 YA(−)

RS250V DI 485
AC

NO

６

CT1

2-M3

35

！

100 to240V AC30

2-φ3.5200！

SAFETY
PRECAUTIONS●●To ensure safe and correct use, thoroughly read and understand the manual before using this instrument.
This instrument is intended to be used for industrial machinery, machine tools and measuring equipment. Verify
correct usage after purpose-of-use consultation with our agency or main oﬃce. (Never use this instrument for
medical purposes with which human lives are involved.)
External protection devices such as protective equipment against excessive temperature rise, etc. must be installed,
as malfunction of this product could result in serious damage to the system or injury to personnel. Proper periodic
maintenance is also required.
This instrument must be used under the conditions and environment described in the manual. Shinko Technos Co.,
Ltd. does not accept liability for any injury, loss of life or damage occurring due to the instrument being used underconditions not otherwise stated in the manual.

Caution with respect toExport Trade Control Ordinance
To avoid this instrument from being used as
a component in, or as being utilized in the
manufacture of weapons of mass destruction(i.e. military applications, military equipment,etc.), please investigate the end users andthe ﬁnal use of this instrument.
In the case of resale, ensure that this
instrument is not illegally exported.

SHINKO TECHNOS CO., LTD.
OVERSEAS DIVISIONUse a solderless terminal with an insulation sleeve in whichan M3 screw ﬁts. Tightening torque should be 0.63 N･m.
φ 3.2mm●●• This catalog is as of March 2018 and its contents are subject to change without notice.
• If you have any inquiries, please consult us or our agency.

■Solderless terminal

5.8 mm max.

−

１

φ12

* To prevent harmful eﬀects due to unexpected level
noise, it is recommended that a surge absorber be
installed between electromagnetic coils.

POWER SUPPLY

5.8 mm max.

＋

24V
AC/DC

40

20

Heater

L
40

Power supply
for alarm unit

25*

■Terminal arrangement
POWER
SUPPLYkk

Plug pin

POWER
SUPPLY

30

100

l
φ5.8

Electromagnetic
switchCTL-12-S36-10L1U (For 50A)

3

ACS-13A-R/M

Horizontal close mounting
n: Number of units mounted！ Caution:

45＋0.50(*) When a terminal cover (sold separately) is used

■Wiring example＋0.5

n×48−3 045＋0.5075

45＋0.50Terminal cover (*)

Gasket A

59.2

Alarm type, setting range and output type, etc. are the same as those of Alarm 1. See Alarm 1 output section.
If this option is ordered, Heating/Cooling control (D□ option) cannot be ordered.
This option and Heater burnout alarm (W, W3 option) utilize common output terminals.
Rating ------------------ Single-phase 20 A [W(20A)], Single-phase 50 A [W(50A)], 3-phase 20 A [W3(20A)], 3-phase 50 A [W3(50A)]
Must be speciﬁed.
Heater burnout alarm Setting range--------- Rated current 20 A [W(20A), W3(20A)]: 0.0 to 20.0 A
Rated current 50 A [W(50A), W3(50A)]: 0.0 to 50.0 A[W, W3 option]
Setting accuracy ---- Within ±5% of the rated value(CT included)
Action ------------------ ON/OFF action
Output ----------------- Relay contact 1a, Control capacity: 3A 250V AC (resistive load), Electrical life: 100,000 cycles
If this option is ordered, Heating/Cooling control (D□ option) cannot be ordered.
Heating control action: Same as the control output (OUT1)
Cooling control action
OUT2 proportional band (P) --- 0.0 to 10.0 times OUT1 proportional band (ON/OFF action when set to 0.0)
OUT2 integral time (I) ------------Same as that of OUT1
OUT2 derivative time (D) ------- Same as that of OUT1
OUT2 proportional cycle -------- 1 to 120 seconds
Heating/Cooling Overlap/Dead band --------------- Thermocouple, RTD: -100.0 to 100.0℃ (°F)
control
Direct current, voltage: -1000 to 1000 (The placement of the decimal point follows the selection.)[D□ option]
OUT2 hysteresis ------------------- 0.1 to 100.0℃ (°F), or 1 to 1000
OUT2 cooling method------------- One cooling method can be selected from Air cooling (linear characteristic), Oil cooling (1.5th power of thelinear characteristic) and Water cooling (2nd power of the linear characteristic) by keypad.
Output ------------------------------- Relay contact 1a, Control capacity: 3A 250V AC (resistive load), Electrical life: 100,000 cycles
Non-contact voltage 12 V DC±15% Max. 40 mA DC (short circuit protected)
If this option is ordered, Alarm 2 (A2 option) and Heater burnout alarm (W, W3 option) cannot be ordered.
Various operations (such as Reading and setting of various set values, Reading of PV and action status and Function change) can be
performed from an external PC.
Communication line ---------------- EIA RS-485
Communication method------------ Half-duplex communication
Synchronization method ---------- Start-stop synchronization
Communication speed ------------ 2400/4800/9600/19200 bps (Selectable by keypad)
Data bit /Parity ---------------------- Data bit: 7 or 8, Parity: Even/Odd /No parity (Selectable by keypad)
Stop bit --------------------------------- 1 or 2 (Selectable by keypad)
Communication protocol --------- Shinko protocol/Modbus ASCII/Modbus RTU (Selectable by keypad)
Number of connectable units --- Max. 31 units per host computer
Serial communication[C5 option]
Communication error detection -- Parity, checksum, LRC (Modbus ASCII), CRC-16 (Modbus RTU)
Data format
Communication protocol
Shinko protocol
Modbus ASCII
Modbus RTU
Start bit1
1
1
Data bit7
7 or 8
8
ParityYes (Even)
Yes (Even, Odd)
Yes (Even, Odd)
No parity
No parity
Stop bit1
1 or 2
1 or 2
If this option is ordered, Set value memory external selection (SM option) cannot be ordered.
SV1, SV2, SV3 or SV4 can be selected by the external contact.
The MEMO display indicates the selected memory number.
Set value
Contact input terminal DI2 can be used for ʻSet value memory external selectionʼ or for ʻOUT/OFF external selectionʼ in [Contact
memory external input function] in Setup mode.
selection
If ʻAuto/Manual control functionʼ is selected in [OUT/OFF key function] in Setup mode, externally Auto/Manual control can be switched.
[SM option]
Circuit current when closed: Approx. 12 mA
If this option is ordered, Serial communication (C5 option) cannot be ordered.
Alarm 2 output[A2 option]

47.5（＊）

When ordering, please specify an option code according to the userʼs needs.

10.5■Optional speciﬁcations

3.2mm

Reg. Oﬃce : 2-5-1, Senbahigashi, Minoo, Osaka, 562-0035, JapanTel: + 81 - 72 - 727 - 6100

Fax

: + 81 - 72 - 727 - 7006

URL

: http://www. shinko-technos. co. jp/e/E-mail: overseas @ shinko-technos. co. jp/e/201803 E06

■External dimensions (Scale: mm)■Panel cutout (Scale: mm)

Mounting frame

44.5
□ 48

7.5

If horizontal close mounting is used for the controller,
IP66 speciﬁcation (Drip-proof/Dust-proof) may
be compromised, and all warranties will be
invalidated.

54.5
59.5（＊）■CT dimensions (Scale: mm)

3-phaseCTL-6-S-H (For 20A)

Power supply
100 to 240V AC or 24V AC/DC

l

15

0.5
7.5

2.8

Alarm unit＋−

K

CTL-6-S

21

10

40

30

Thermocouple
Electric
furnace

２

１

１３

２ 3A
３

EV1

250V
AC

NO

４ 3A
＋
O2/EV2

−

５

CT2

＋
O1

１４ −
１５
１６
１７

DI1 YB(＋)
COM SG

７

3A
250V
AC

15

■Console software SWS-ACS01M included

Using the Console software (SWS-ACS01M) with an USB communication
cable CMA (sold separately), parameters setting, logging and monitoring of
the controller can be performed by connecting to the USB port of the PC.

■USB communication cable CMA(Sold Separately) (Scale: mm)

USB plug

８

９

＋＋ A

１０

A
V

１１
B

１８ −−− B１２
DC TC RTD

Alarm 1 output

O2/EV2

Cooling output (D□ option), Alarm 2 output (A2 Option)
or Heater burnout alarm output (W, W3 option)

O1

Control output or Heating output (D□ option)

DC

Direct current or DC voltage input (For DC voltage input,+ side terminal number diﬀers depending on the voltage input.)TC

Thermocouple input

RTD

RTD input

CT1

CT input 1 (W, W3 option)

CT2

CT input 2 (W3 option)

DI

Contact input (SM option)

RS-485

Serial communication RS-485 (C5 option)！ ・This controller does not have a built-in power switch, circuit breaker and fuse.

It is necessary to install a power switch, circuit breaker and fuse near the controller.
・For a 24 V AC/DC power source, do not confuse polarity when using
direct current (DC).

50
2000

EV1

NO

＋

＞1V

DI2 YA(−)

RS250V DI 485
AC

NO

６

CT1

2-M3

35

！

100 to240V AC30

2-φ3.5200！

SAFETY
PRECAUTIONS●●To ensure safe and correct use, thoroughly read and understand the manual before using this instrument.
This instrument is intended to be used for industrial machinery, machine tools and measuring equipment. Verify
correct usage after purpose-of-use consultation with our agency or main oﬃce. (Never use this instrument for
medical purposes with which human lives are involved.)
External protection devices such as protective equipment against excessive temperature rise, etc. must be installed,
as malfunction of this product could result in serious damage to the system or injury to personnel. Proper periodic
maintenance is also required.
This instrument must be used under the conditions and environment described in the manual. Shinko Technos Co.,
Ltd. does not accept liability for any injury, loss of life or damage occurring due to the instrument being used underconditions not otherwise stated in the manual.

Caution with respect toExport Trade Control Ordinance
To avoid this instrument from being used as
a component in, or as being utilized in the
manufacture of weapons of mass destruction(i.e. military applications, military equipment,etc.), please investigate the end users andthe ﬁnal use of this instrument.
In the case of resale, ensure that this
instrument is not illegally exported.

SHINKO TECHNOS CO., LTD.
OVERSEAS DIVISIONUse a solderless terminal with an insulation sleeve in whichan M3 screw ﬁts. Tightening torque should be 0.63 N･m.
φ 3.2mm●●• This catalog is as of March 2018 and its contents are subject to change without notice.
• If you have any inquiries, please consult us or our agency.

■Solderless terminal

5.8 mm max.

−

１

φ12

* To prevent harmful eﬀects due to unexpected level
noise, it is recommended that a surge absorber be
installed between electromagnetic coils.

POWER SUPPLY

5.8 mm max.

＋

24V
AC/DC

40

20

Heater

L
40

Power supply
for alarm unit

25*

■Terminal arrangement
POWER
SUPPLYkk

Plug pin

POWER
SUPPLY

30

100

l
φ5.8

Electromagnetic
switchCTL-12-S36-10L1U (For 50A)

3

ACS-13A-R/M

Horizontal close mounting
n: Number of units mounted！ Caution:

45＋0.50(*) When a terminal cover (sold separately) is used

■Wiring example＋0.5

n×48−3 045＋0.5075

45＋0.50Terminal cover (*)

Gasket A

59.2

Alarm type, setting range and output type, etc. are the same as those of Alarm 1. See Alarm 1 output section.
If this option is ordered, Heating/Cooling control (D□ option) cannot be ordered.
This option and Heater burnout alarm (W, W3 option) utilize common output terminals.
Rating ------------------ Single-phase 20 A [W(20A)], Single-phase 50 A [W(50A)], 3-phase 20 A [W3(20A)], 3-phase 50 A [W3(50A)]
Must be speciﬁed.
Heater burnout alarm Setting range--------- Rated current 20 A [W(20A), W3(20A)]: 0.0 to 20.0 A
Rated current 50 A [W(50A), W3(50A)]: 0.0 to 50.0 A[W, W3 option]
Setting accuracy ---- Within ±5% of the rated value(CT included)
Action ------------------ ON/OFF action
Output ----------------- Relay contact 1a, Control capacity: 3A 250V AC (resistive load), Electrical life: 100,000 cycles
If this option is ordered, Heating/Cooling control (D□ option) cannot be ordered.
Heating control action: Same as the control output (OUT1)
Cooling control action
OUT2 proportional band (P) --- 0.0 to 10.0 times OUT1 proportional band (ON/OFF action when set to 0.0)
OUT2 integral time (I) ------------Same as that of OUT1
OUT2 derivative time (D) ------- Same as that of OUT1
OUT2 proportional cycle -------- 1 to 120 seconds
Heating/Cooling Overlap/Dead band --------------- Thermocouple, RTD: -100.0 to 100.0℃ (°F)
control
Direct current, voltage: -1000 to 1000 (The placement of the decimal point follows the selection.)[D□ option]
OUT2 hysteresis ------------------- 0.1 to 100.0℃ (°F), or 1 to 1000
OUT2 cooling method------------- One cooling method can be selected from Air cooling (linear characteristic), Oil cooling (1.5th power of thelinear characteristic) and Water cooling (2nd power of the linear characteristic) by keypad.
Output ------------------------------- Relay contact 1a, Control capacity: 3A 250V AC (resistive load), Electrical life: 100,000 cycles
Non-contact voltage 12 V DC±15% Max. 40 mA DC (short circuit protected)
If this option is ordered, Alarm 2 (A2 option) and Heater burnout alarm (W, W3 option) cannot be ordered.
Various operations (such as Reading and setting of various set values, Reading of PV and action status and Function change) can be
performed from an external PC.
Communication line ---------------- EIA RS-485
Communication method------------ Half-duplex communication
Synchronization method ---------- Start-stop synchronization
Communication speed ------------ 2400/4800/9600/19200 bps (Selectable by keypad)
Data bit /Parity ---------------------- Data bit: 7 or 8, Parity: Even/Odd /No parity (Selectable by keypad)
Stop bit --------------------------------- 1 or 2 (Selectable by keypad)
Communication protocol --------- Shinko protocol/Modbus ASCII/Modbus RTU (Selectable by keypad)
Number of connectable units --- Max. 31 units per host computer
Serial communication[C5 option]
Communication error detection -- Parity, checksum, LRC (Modbus ASCII), CRC-16 (Modbus RTU)
Data format
Communication protocol
Shinko protocol
Modbus ASCII
Modbus RTU
Start bit1
1
1
Data bit7
7 or 8
8
ParityYes (Even)
Yes (Even, Odd)
Yes (Even, Odd)
No parity
No parity
Stop bit1
1 or 2
1 or 2
If this option is ordered, Set value memory external selection (SM option) cannot be ordered.
SV1, SV2, SV3 or SV4 can be selected by the external contact.
The MEMO display indicates the selected memory number.
Set value
Contact input terminal DI2 can be used for ʻSet value memory external selectionʼ or for ʻOUT/OFF external selectionʼ in [Contact
memory external input function] in Setup mode.
selection
If ʻAuto/Manual control functionʼ is selected in [OUT/OFF key function] in Setup mode, externally Auto/Manual control can be switched.
[SM option]
Circuit current when closed: Approx. 12 mA
If this option is ordered, Serial communication (C5 option) cannot be ordered.
Alarm 2 output[A2 option]

47.5（＊）

When ordering, please specify an option code according to the userʼs needs.

10.5■Optional speciﬁcations

3.2mm

Reg. Oﬃce : 2-5-1, Senbahigashi, Minoo, Osaka, 562-0035, JapanTel: + 81 - 72 - 727 - 6100

Fax

: + 81 - 72 - 727 - 7006

URL

: http://www. shinko-technos. co. jp/e/E-mail: overseas @ shinko-technos. co. jp/e/201803 E06