SC series

AC Transducer

RMS computation

3rd harmonic wave
Low cost, Light,Space saving

Confirms toJIS C1111

Universalpower supply(Actual size)

AC voltage

SCPT

AC current

SCCT(AC/DC)

Both DIN rail and panel
mounting usable

TerminalcoverprovidedModelAC voltage transducer (W25×H100×D128mm)
0 to 150V AC
0 to 300V AC
0 to 86.6V AC
0 to 110V AC
4 to 20mA DC (Load current 600Ω or less)0 to 1mA DC (Load current 10kΩ or less)
1 to 5V DC (Load current 1kΩ or less)0 to 5V DC (Load current 1kΩ or less)0 to 10V DC (Load current 1kΩ or less)
Universal: 85 to 264V AC , 85 to 143V DC
20～30V DC
40～60V DC

Input

OutputPower supply

Input

OutputPower supply

AC current transducer (W25×H100×D128mm)
0 to 5A AC
0 to 1A AC
0 to 6A AC
4 to 20mA DC (Load current 600Ω or less)0 to 1mA DC (Load current 10kΩ or less)
1 to 5V DC (Load current 1kΩ or less)0 to 5V DC (Load current 1kΩ or less)0 to 10V DC (Load current 1kΩ or less)
Universal: 85 to 264V AC , 85 to 143V DC
20 to 30V DC40 to 60V DCSpecifications
External dimensions
Mounting
Case, Terminal block
Conversion accuracy
Output ripple
Response timeContinuous load
Momentary overload
Insulation resistanceDielectric strength
Power supply
Power consumption
Ambient temperature
Ambient humidity
WeightAccessories

25×100×128mm（W×H×D）
Both DIN rail and panel
Flame-resistant resin Color: Black (Teminal cover: Transparent polycarbonate)±0.5% of output span (Conditions: Ambient temperature 23±2℃, Relative Humidity 45 to 70%RH)±1% P-P of output span
Within 0.5 seconds (90% step input response)1.2 times of rated input current
SCPT: Twice of rated input voltage (for 10sec)
SCCT: 10 times of rated input current (for 16sec), 20 times of rated input crrent (for 4sec), 40 times of rated input current (for 1sec)
Input/Output/Power-Ground, Input-Output, Input/Output-Power (SCPT: Output-ground) 50MΩ or more, at 500V DC
Input/Output/Power-Ground, Input-Output, Input/Output-Power (SCPT: Output-ground) 2kV AC for 1 minute85 to 264V AC 50/60Hz, 85 to 143V DC
Measuring side: 0.3VA or less Power side: 2VA or less Universal power supply (24/48V DC: 2W or less)-10 to 55℃
30 to 85%RH (No condensation)
Approx. 200g
instruction manual 1 copy, Terminal cover 1 piece

Terminal arrangement

PT (insulated current transformer) and CT(current trensformer) are sold separately.
For more information, please contact ourmain office or dealers.

External dimensions (Unit: mm)

Mounting
DIN rail mouting

Panel mounting

For mounting, hook the unti on theupper side of the DIN raik top to bottom.

● To ensure safe and correct use, thoroughly read and understand the manual before using this instrument.
● This instrument is intended to be used for industrial machinery, machine tools and measuring equipment. Verify
correct usage after purpose-of-use consultation with our agency or main office.
(Never use this instrument for medical purposes with which human lives are involved.)● External protection devices such as protection equipment against excessive temperature rise, etc. must be installed,
as malfunction of this product could result in serious damage to the system or injury to personnel. Also proper
periodic maintenance is required.
● This instrument must be used under the conditions and environment described in the manual. Shinko Technos Co.,
Ltd. does not accept liability for any injury, loss of life or damage occurring due to the instrument being used underconditions not otherwise stated in this manual.

M4 screws can also
be used for mountinng

To avoid this instrument from being used as acomponent in, or as being utilized in themanufacture of weapons of mass destruction(i.e. military applications, military equipment,etc.), please investigate the end users andthe ﬁnal use of this instrument.
In the case of resale, ensure that thisinstrument is not illegally exported.

・ This catalog is as of June 2021, and specifications are subject to change without notice.
・ Photos used in this catalog do not show unit in operating status.
・ If you have any inquiries, please consult us or our agency.

SHINKO TECHNOS CO., LTD.
OVERSEAS DIVISION

Head Office : 2-5-1, Senbahigashi, Minoo, Osaka, 562-0035, JapanTel: +81-72-727-6100
Fax: +81-72-727-7006
URL: https://shinko-technos.co.jp/e/E-mail: overseas@shinko-technos.co.jp
202106 E04