SE series

SPEC. SHEET

Current Loop Supply(with indication function)

Model:SE1D■ ModelSE1D - □ - □Socket
1: Finger-safe(For Y terminal)
2: For Ring terminal
Power supply
0: 100 to 240V AC
1: 24V AC/DC■ How to orderSpecify the model (e.g.) SE1D-1-0
Default value
Input
4 to 20mA DC
Output4 to 20mA DC■ Performance■ Accessories (sold separately)
Communication cable for the console software: CMB-001■ Input specification
DC current
Connect a shunt resistor (sold separately) between Shunt resistor

Input
4 to 20mA DC

Shunt resistor
50 built-in■ Output specification
When the output range lower limit is zero, (even if zeroadjustment results in a negative value), the output value
will not be negative.

Accuracy (When ambient temperature is 23 ):Input (DC current): Within 0.1%
Output: Within 0.1%
Indication accuracy: Within input accuracy 1 digit
Input sampling period: 25ms, 125ms, 250ms (Selectable by keypad)
Response time: 65ms (typ.) (0→90%) (Input sampling period 25ms)
225ms (typ.) (0→90%) (Input sampling period 125ms)
425ms (typ.) (0→90%) (Input sampling period 250ms)(Selectable by keypad)
Temperature coefficient: 0.015%/ or lessTemperature coefficient: 0.015%/ or lessInsulation resistance: 10M or more, at 500V DC(Input – Output – Power supply)
Dielectric strength: 2.0kV AC for 1 minute(Input – Output – Power supply)■ General structure

DC Current
Output range

Allowable loadresistance

4 to 20mA DC
0 to 20mA DC
0 to 12mA DC
0 to10mA DC
1 to 5mA DC

700
700
1.2k
1.2k
2.4k

or lessor lessor lessor lessor less

Zeroadjustmentrange-5 to 5%0 to 5%0 to 5%0 to 5%-5 to 5%Span
adjustment
range
95 to 105%
95 to 105%
95 to 105%
95 to 105%
95 to 105%

Zeroadjustment
range
0 to 5%0 to 5%-5 to 5%0 to 5%Span
adjustment
range
95 to 105%
95 to 105%
95 to 105%
95 to 105%

DC Voltage
Output range

Allowable loadresistance

0 to 1V DC
0 to 5V DC
1 to 5V DC
0 to 10V DC

100 or more
500 or more
500 or more
1k or moreCase: Frame-resistant resin, Color: Light gray
Front panel: Membrane sheet
Setting: By the front keypad
Connector for console software: Only for CMB-001
Indication: Input display: 7-segment, Red LED display 4-digit
Character size 10×4.6mm (H×W)
Output display: 7-segment, Red LED display 4-digit
Character size 10×4.6mm (H×W)
Power indicator: Green LED■ Power for 2-wire transmitter
Output voltage: 24 to 28V DC (When load current is 20mA)
Ripple voltage: Within 200mV DC (When load current is 20mA)
Max load current: 25mA DC
Console connector■ Installation specifications

SE1DmA DC

Power supply: 100 to 240V AC 50/60Hz, 24V AC/DC 50/60Hz■ Circuit configuration and terminal arrangement
Allowable voltage range: 85 to 264V AC, 20 to 28V AC/DC+ 1 50Ω
+Power consumption : Approx. 8VA
13Input
PowerDisplay/Setting
circuitPower supply
Ambient temperature: -5 to 55
circuitkey circuit
214
Ambient humidity: 35 to 85%RH (Non-condensing)3
Insulation circuit
Power supply for+2-wire transmitter
Mounting: DIN rail mounting
924V DC
OutputCPUOutput
4Powercircuit+
External dimensions: W30×H88×D108mm (including the socket)
circuit
105Weight: Approx. 180g (including the socket)
6117■ Attached functions
Auto-light function: Display brightness is controlled in accordancewith the surrounding area. Unnecessary brightness is reduced,saving energy.
Power failure countermeasure:The data is backed up in non-volatile IC memory.
Self diagnosis:
The CPU is monitored by a watchdog timer, and when anabnormal status is found on the CPU, the unit is switched towarm-up status with tuning all outputs off.

128

When using as a Current loop supply-+ 1- 22-wiretransmitter

3++4

When using as an Isolator+50Ω
Input
circuitInput-1250Ω
Inputcircuit

324V DCpowercircuit

424V DCpowercircuit■ Settings
Function keys
(1) UP Key: Increases the numeric value.
(2) DOWN Key: Decrease the numeric value.
(3) MODE Key: Selects the setting mode.
(4) SUB-MODE Key: Turns the displays ON again when theyare in OFF status.

■ External dimensions (Scale: mm)(The UP, DOWN or MODE Key also turns the displays ON again
when they are in OFF status.)■ Displays and indicators
Input display: Indicates the input value
Indication of -200.0 or less (for the range withdecimal point):
The minus (-) sign and input value light alternately.
Indication of -2000 or less (for DC input):
The minus (-) sign and input value light alternately.
Indication of 10000 or more:
The lower 4 digits flash.
Under range: “” flashes on the input display.
Over range: “” flashes on the input display
Warm-up indication: For approx. 3sec. after the power to theinstrument is turned on, the input type isindicated on the input display, and Output
type is indicated on the Output display.
Output display: Indicates output volume in percentage (%) form.
Power indicator: The green LED lights when the power to the
instrument is turned on.

■ Solderless terminal
Y typeRing type

SHINKO TECHNOS CO., LTD.

Tel: 81-72-727-6100

URL: http://www.shinko-technos.co.jp/e/ E-mail: overseas@shinko-technos.co.jp

20120106