SE seriesIsolator■ ModelSPEC. SHEET(with indication function)

Model:SE1V

SE1V - □ - □Socket
1: Finger-safe(For Y terminal)
2: For Ring terminal
Power supply
0: 100 to 240V AC
1: 24V AC/DC■ How to orderSpecify the model (e.g.) SE1V-1-0
Default value
Input
1 to 5V DC
Output4 to 20mA DC■ Performance■ Accessories (sold separately)
Communication cable for the console software: CMB-001■ Input specification
DC voltage
Input

Input resistance

0 to 10mV DC-10 to 10mV DC
0 to 50mV DC
0 to 60mV DC
0 to 100mV DC0 to 1V DC
0 to 5V DC
1 to 5V DC
0 to 10V DC

1MAllowable signal
source resistance
20 or less40 or less200or less2kor less1kor less■ Output specification
When the output range lower limit is zero, (even if zeroadjustment results in a negative value), the output value
will not be negative.
DC current
ZeroSpan
Allowable load
Output range
adjustment
adjustment
resistance
range
range
4 to 20mA DC
700 or less
-5 to 5%95 to 105%
0 to 20mA DC
700 or less0 to 5%95 to 105%
0 to 12mA DC
1.2k or less0 to 5%95 to 105%
0 to10mA DC
1.2k or less0 to 5%95 to 105%
1 to 5mA DC
2.4k or less
-5 to 5%95 to 105%

Accuracy (When ambient temperature is 23 ):Input: Within 0.1%
Output: Within 0.1%
Indication accuracy: Within input accuracy 1 digit
Input sampling period: 25ms, 125ms, 250ms (Selectable by keypad)
Response time: 65ms (typ.) (0→90%) (Input sampling period 25ms)
225ms (typ.) (0→90%) (Input sampling period 125ms)
425ms (typ.) (0→90%) (Input sampling period 250ms)(Selectable by keypad)
Temperature coefficient: 0.015%/ or lessInsulation resistance: 10M or more, at 500V DC(Input – Output – Power supply)
Dielectric strength: 2.0kV AC for 1 minute(Input – Output – Power supply)■ General structure
Case: Flame-resistant resin, Color: Light gray
Front panel: Membrane sheet
Setting: By the front keypad
Connector for console software: Only for CMB-001
Indication: Input display: 7-segment, Red LED display 4-digit
Character size 10×4.6mm (H×W)
Output display: 7-segment, Red LED display 4-digit
Character size 10×4.6mm (H×W)
Power indicator: Green LED

DC voltage
Output range

Allowable loadresistance

0 to 1V DC
0 to 5V DC
1 to 5V DC
0 to 10V DC

100 or more
500 or more
500 or more
1k or more

Zeroadjustment
range
0 to 5%0 to 5%-5 to 5%0 to 5%Span
adjustment
range
95 to 105%
95 to 105%
95 to 105%
95 to 105%

Console connector

SE1V■ Solderless terminal■ Installation specifications

Y typePower supply: 100 to 240V AC 50/60Hz, 24V AC/DC 50/60Hz
Allowable voltage range: 85 to 264V AC, 20 to 28V AC/DCPower consumption: Approx. 6VA
Ambient temperature: -5 to 55
Ambient humidity: 35 to 85%RH (Non-condensing)
Mounting: DIN rail mounting
External dimensions: W30×H88×D108mm (including the socket)
Weight: Approx. 190g (including the socket)■ Attached functions
V( 10V)■ Circuit configuration and terminal arrangementmV( 1V)

Auto-light function: Display brightness is controlled in accordancewith the surrounding area. Unnecessary brightness is reduced,saving energy.
Power failure countermeasure: The data is backed up in nonvolatile IC memory.
Self diagnosis: The CPU is monitored by a watchdog timer, andwhen an abnormal status is found on the CPU, the unit isswitched to warm-up status with tuning all outputs off.

Ring type+--

123+

Input
circuit

Display/Setting
key circuit

CPU5RoHS directive compliance

613+

14-

9+

10-

Power
supplyInsulation circuit

4■ Environmental specificationPower
circuitOutputcircuit

78Output

1112■ Settings
Function keys
(1) UP Key: Increases the numeric value,(2) DOWN Key: Decrease the numeric value,(3) MODE Key: Selects the setting mode,(4) SUB-MODE Key: Turns the displays ON again when theyare in OFF status.

■ External dimensions (Scale: mm)(The UP, DOWN or MODE Key also turns the displays ON again
when they are in OFF status,)■ Displays and indicators
Input display: Indicates the input value
Indication of -2000 or less (for DC input):
The minus (-) sign and input value light alternately.
Indication of 10000 or more:
The lower 4 digits flash.
Under range: “” flashes on the input display.
Over range: “” flashes on the input display
Warm-up indication: For approx. 3sec. after the power to theinstrument is turned on, the input type isindicated on the input display, and Output
type is indicated on the Output display.
Output display: Indicates output volume in percentage (%) form.
Power indicator: The green LED lights when the power to the
instrument is turned on.

SHINKO TECHNOS CO., LTD.

Tel: 81-72-727-6100

URL: http://www.shinko-technos.co.jp/e/ E-mail: overseas@shinko-technos.co.jp

20120106