SE series

SPEC. SHEET

Universal Transmitter

Model:SE1U(with indication function)■ ModelSE1U - □ - □Socket
1: Finger-safe(For Y terminal)
2: For Ring terminal
Power supply
0: 100 to 240V AC
1: 24V AC/DC■ How to orderSpecify the model (e.g.) SE1U-1-0
Default value
Input
1 to 5V DC
Output4 to 20mA DC■ Accessories (sold separately)
Model
Spec.
RES-S01-050
500.1%Ring
RES-S01-100
1000.1%terminal RES-S01-200
2000.1%Shunt
RES-S01-01K
1k0.1%
resistor
RES-S06-050
500.1%RES-S06-100Y1000.1%
terminal RES-S06-200
2000.1%RES-S06-01K
1k0.1%
Communication cable for the console software: CMB-001

DC current
Connect a shunt resistor (sold separately) between input terminals

Input
4 to 20mA DC
0 to 20mA DC
0 to 16mA DC
2 to 10mA DC
0 to 10mA DC
1 to 5mA DC
0 to 1mA DC

Name■ Input specification
Thermocouple
Input resistance: 1M or more
External resister: 100 or less, however, B input, 40 or less
Burnout: Upscale, Downscale (Selectable by keypad)
Input
Thermocouple
Input range
K-200 to 1370
-328 to 2498
J-200 to 1000
-328 to 1832
R-50 to 1760
-58 to 3200
S-50 to 1760
-58 to 3200
B
0 to 1820
32 to 3308
E-200 to 800-328 to 1472
T-200 to 400-328 to 752
N-200 to 1300
-328 to 2372
PL0 to 1390
32 to 2534
W5Re/W26Re
0 to 2315
32 to 4199
W3Re/W25Re
0 to 2315
32 to 4199
Minimum span: 50 (100 )RTD (3-wire system)
Input detection current: Approx. 0.2mA
Allowable lead wire resistance: 10 or less per wire
Burnout: Upscale, Downscale (Selectable by keypad)
Input
RTD
Input range
Pt100-200 to 850-328 to 1562
JPt100-200 to 500-328 to 932
Minimum span: 50 (100 )Shunt resistor
50100200
1k

DC voltage
Input

Input resistance

0 to 10mV DC-10 to 10mV DC
0 to 50mV DC
0 to 60mV DC
0 to 100mV DC
0 to 1V DC
0 to 5V DC
1 to 5V DC
0 to 10V DC

1MAllowable signal
source resistance
20 or less40 or less200or less2kor less1kor less■ Output specification
When the output range lower limit is zero, (even if zeroadjustment results in a negative value), the output value will not
be negative.
DC Current
Zero
Span
Allowable load
Output range
adjustment
adjustment
resistance
range
range
4 to 20mA DC
700 or less
-5 to 5%95 to 105%
0 to 20mA DC
700 or less0 to 5%95 to 105%
0 to 12mA DC
1.2k or less0 to 5%95 to 105%
0 to10mA DC
1.2k or less0 to 5%95 to 105%
1 to 5mA DC
2.4k or less
-5 to 5%95 to 105%
DC Voltage
Zero
Span
Allowable load
Output range
adjustment
adjustment
resistance
range
range
0 to 1V DC
100 or more
0 to 5%95 to 105%
0 to 5V DC500 or more
0 to 5%95 to 105%
1 to 5V DC
500 or more
-5 to 5%95 to 105%
0 to 10V DC
1k or more
0 to 5%95 to 105%

SE1U
■ Performance■ Environmental specification

Accuracy (When ambient temperature is 23 ):Input:･ Thermocouple:
Within 0.1% of each input span, however,R, S input, -50 to 200 (-58 to 392 ): Within 6 (12 )B input, 0 to 300 (32 to 572 ): Accuracy is not guaranteed.
K, J, E, T, N input, less than 0 (32 ):Within 0.4% of each input span
･ RTD: Within 0.1% of each input span･ DC current: Within 0.1%
･ DC voltage: Within 0.1%
Output: Within 0.1%
Cold junction compensation accuracy: Within 1 at -5 to 55
Indication accuracy: Within input accuracy 1 digit
Input sampling period: 25ms, 125ms, 250ms (Selectable by keypad)
Response time: 65ms (typ.) (0→90%) (Input sampling period 25ms)
225ms (typ.) (0→90%) (Input sampling period 125ms)
425ms (typ.) (0→90%) (Input sampling period 250ms)(Selectable by keypad)
Temperature coefficient: 0.015%/ or lessInsulation resistance: 10M or more, at 500V DC(Input – Output – Power supply)
Dielectric strength: 2.0kV AC for 1 minute(Input – Output – Power supply)

RoHS directive compliance■ General structure
Case: Flame-resistant resin, Color: Light gray
Front panel: Membrane sheet
Setting: By the front keypad
Connector for console software: Only for CMB-001
Indication: Input display: 7-segment, Red LED display 4-digit
Character size 10×4.6mm (H×W)
Output display: 7-segment, Red LED display 4-digit
Character size 10×4.6mm (H×W)
Power indicator: Green LED■ Settings
Function keys(1) UP Key: Increases the numeric value,(2) DOWN Key: Decrease the numeric value,(3) MODE Key: Selects the setting mode,(4) SUB-MODE Key: Turns the displays ON again when theyare in OFF status.
(The UP, DOWN or MODE Key also turns the displays ON again
when they are in OFF status,)■ Displays and indicators
Input display: Indicates the input value
Indication of -200.0 or less (for the range withdecimal point):
The minus (-) sign and input value light alternately.
Indication of -2000 or less (for DC input):
The minus (-) sign and input value light alternately.
Indication of 10000 or more:
The lower 4 digits flash.
Under range: “” flashes on the input display.
Over range: “” flashes on the input display
Warm-up indication: For approx. 3sec. after the power to the
instrument is turned on, the input type isindicated on the input display, and Output
type is indicated on the Output display.
Output display: Indicates output volume in percentage (%) form.
Power indicator: The green LED lights when the power to the
instrument is turned on.

■ Solderless terminal
Y typeRing type

■ Circuit configuration and terminal arrangement

Console connector■ Installation specifications
Power supply: 100 to 240V AC 50/60Hz, 24V AC/DC 50/60Hz
Allowable voltage range: 85 to 264V AC, 20 to 28V AC/DCPower consumption: Approx. 6VA
Ambient temperature: -5 to 55
Ambient humidity: 35 to 85%RH (Non-condensing)
Mounting: DIN rail mounting
External dimensions: W30×H88×D108mm (including the socket)
Weight: Approx. 190g (including the socket)■ External dimensions (Scale: mm)■ Attached functions
Auto-light function: Display brightness is controlled in accordancewith the surrounding area. Unnecessary brightness is reduced,saving energy.
Power failure countermeasure: The data is backed up in nonvolatile IC memory.
Self diagnosis: The CPU is monitored by a watchdog timer, andwhen an abnormal status is found on the CPU, the unit isswitched to warm-up status with tuning all outputs off.
Cold junction compensation: Available

SHINKO TECHNOS CO., LTD.

Tel: 81-72-727-6100

URL: http://www.shinko-technos.co.jp/e/ E-mail: overseas@shinko-technos.co.jp

20120106