Digital Indicating Controllers
ACD-13A,ACR-13A

ON/OFF SERVO Digital Indicating Controllers
ACD-15A,ACR-15A

Visibility,
Functionality

Simpliﬁed setting – Set frequently used settings for streamline

96

96

96

Easy status checking using 3-color switching

110

48

Industry leading large display

Easier viewing display

110Industry Leading Large Display

Multi-Functions, Simple to Operate

PID zone function: PID resetting due to SV change Unnecessary

Large LCD display

A specially treated large LCD display makes it easier to view evenin bright light and open-air.
PV display (ACD series): 24.0 x 11.0mm (H x W)

Up to 5 groups of PID parameters can be set.
When SV is changed, PID parameters are automatically changed for optimal
control. (It is not necessary to reset PID after SV is changed.)

Digital Indicating Controllers
ACD-13A, ACR-13A

SV (desired value)

ON/OFF SERVO Digital Indicating Controllers

Z o n e①

ACD-15A, ACR-15A

Z o n e②
Setting PID Zone to 300℃

250℃

JCD- 33 A

ACD series

WhenchangedAn easily viewable bar graph

100℃

・(e.g.) MV 50%-5%・(e.g.) MV 100%
-5%

105%

・(e.g.) Deviation 0 (zero)-0

Without setting engineering items, simplified setting modecan prevent operational mistakes, and simple operations
run smoothly. Basic settings and key operations now aredoable via 3-key usage.

+

・(e.g.) Negative deviation-

0+Power unnecessary if USB comm. cable used

Enhanced visibility

PV display color selectable from red, green and orange.
Colors can be set depending on the deviation between
PV and SV, so status can be checked from a distance.
● PV color continuous change mode
When PV is higher than
the setting range

When PV is within thesetting range● It is easier to see the SV, PV and setting characters,
as an 11-segment LCD display is used.

All segments
are backlit

SENS

By connecting to a PC using the USB communication
cable (CMB-001) (sold separately), initial settings can bemade easily without complicated wiring.
[USB communication cable (sold separately) anddedicated software (free of charge) are required.]

Actual size

OS: Windows 7/8/10 (Japanese/English)
https://shinko-technos.co.jp/e/ → Support & Downloads → Downloads →Software → ACD, ACR-13A(15A) console software (SWS-AC001M)

ARW● PV display color is selectable from 7 modes below.
When PV is lower than
the setting range

If SV is changed from 100℃to
250℃, PID zone ( higher than200℃, lower than 300℃) is
automatically changed.

Simple operation in Simpliﬁed setting mode

In the case of zero (0) deviation, central 2bars light.
For positive deviation, bars light increasingly
to the right.
For negative deviation, bars light
Increasingly to the left.

105%

Setting PID Zone to 200℃

Z o n e④

PID Zone changes
from ④ to ③

DV indication

Scale is -5 to 105%, andbars light increasingly to theright in accordance with the MV.

Z o n e③Z o n e⑤22-segment bar graph allows simultaneous PV,
SV, MV viewing.
Ease of viewing for manual output operation.
For the ACD-15A and ACR-15A, the motor valve
opening can be checked with the bar graph.
(when feedback potentiometer “Yes” is set)
MV indication

PID zone setting• PV display: Green, Red, Orange• Event output (any event from EVT1 to EVT5)
Alarm OFF: Green, Alarm ON: Red
Alarm OFF: Orange, Alarm ON: Red• PV color changes continuously : Orange→Green→Red• PV color changes continuously + Event output (any event from EVT1 to
EVT5) ON (Red)USB comm. cable (CMB-001)
ACR-13A bottom

Feedback potentiometer“Yes / No” selectable

PC

Selectable using the front keypad. If “Yes” is selected, feedback potentiometer
position Fully Closed/Fully Open can be automatically adjusted. If “No” isselected, it is manually adjustable (only for ACD-15A, ACR-15A).

■ Model■ Rated scale
Input

ACD - 1 □ A - □ / M □ , □ □ □ W96×H96mm
ACR - 1 □ A - □ / M □ , □ □ □ W48×H96mm
Controlaction3
5

A1
Control output(OUT1)

A
Ｒ
Ｓ
A

Input
Supply voltage

Ｍ
1
EI
A3
A5
W
W3
DR
DS

Option(Multiple options selectable)

DA
C
C5
EA1
EA2
EV1
EV2
TA1
TV1
P

PID
ON/OFF servo output PID
Selectable with the keypad operation (*1)
Relay contact :1a1b (ACx-15A: 1a x 2)
Non-contact voltage(SSR drive): 12VDC ±15%
DC current: 4 to 20mA DC
Multi-range (*2)
100 to 240V AC(Standard)
24V AC/DC (*3)
Event input (*5)
Event output (EVT1 to EVT3) (*4)(*6)(*7)
Event output (EVT4, EVT5)Heater burnout
Single-phasealarm (*4)(*8)
3-phaseRelay contact: 1a
Non-contact voltage (SSR Heating / Cooling
control output
drive):12V DC±15%(OUT2) (*4)(*6)(*7)
DC current:
4 to 20mA DC
Serial
RS-232C
communication (*5)
RS-4854 to 20mA DC
External setting
0 to 20mA DC
input
0 to 1V DC
1 to 5V DC
Transmission
4 to 20mA DC (*9)
output0 to 1V DC (*10)
Insulated power output 24V DC (*4)(*6)(*7)

Scale range
K

Thermocouple-200 to 1370 ℃-200.0 to 400.0 ℃-328 to 2498 °Ｆ

Resolution
1℃ (°
Ｆ)

Ｆ 0.1℃ (°Ｆ)-328.0 to 752.0 °J-200 to 1000 ℃-328 to 1832 °
Ｆ

1℃ (°
Ｆ)

R

0 to 1760 ℃

Ｆ
32 to 3200 °

Ｆ)
1℃ (°

S

0 to 1760 ℃

Ｆ
32 to 3200 °

Ｆ)
1℃ (°

B

0 to 1820 ℃

Ｆ
32 to 3308 °

Ｆ)
1℃ (°

E

-200 to 800 ℃-328 to 1472 °
Ｆ

1℃ (°
Ｆ)

T

-200.0 to 400.0℃

N-200 to 1300 ℃

Ｆ-328 to 2372 °

Ｆ)
1℃ (°

0 to 1390 ℃

Ｆ
32 to 2534 °

Ｆ)
1℃ (°

0 to 2315 ℃

32 to 4199 °
Ｆ

1℃ (°
Ｆ)

PL-Ⅱ
C(W/Re5-26)-328.0 to 752.0 °Ｆ 0.1℃ (°Ｆ)

Ｆ 0.1℃ (°Ｆ)-200.0 to 850.0℃ -328.0 to 1562.0 °
Pt100

RTD

-148.0 to 212.0 °Ｆ 0.1℃ (°Ｆ)
Ｆ 0.1℃ (°Ｆ)-148.0 to 932.0 °-200 to 850 ℃
JPt100DC current-100.0 to 100.0℃-100.0 to 500.0 ℃

4 to 20mA-200.0 to 500.0℃-200 to 500 ℃-328 to 1562 °
Ｆ

1℃ (°
Ｆ)

-328.0 to 932.0 °Ｆ 0.1℃ (°Ｆ)
Ｆ-328 to 932 °

Ｆ)
1℃ (°

0 to 20mA
0 to 10mV-10 to 10mV
0 to 50mV

DC voltage

0 to 100mV-2000 to 10000 *1

1

0 to 1V
0 to 5V
1 to 5V
0 to 10V*1: Decimal point place change and scaling are possible.
(*1): Alarm types (12 types and No alarm action) and status Energized/De-energized can be
set by front keypad.
(*2): Thermocouple, RTD, DC current or DC voltage is selectable by front keypad.
(*3): For the supply voltage, 100 to 240V AC is standard.
When ordering 24V AC/DC, enter “1” after the input code.
(*4): Applicable to the ACD-13A, ACR-13A.
(*5): If EI and C/C5 options are added together, Event input EVI3 and EVI4 cannot be used.
(*6): A3, D□ and P options cannot be added together.
(*7): If D□ and P options are added, Event output EVT2 cannot be used.
(*8): The rated current (20A, 100A) for single phase and 3-phase is selectable by front
keypad. The CT is sold separately. Not available for the DC current output type.
(*9): It can be changed to "0 to 20 mA DC", "0 to 5 mA DC" or "0 to 12 mA DC". Please consult us.
(*10): It can be changed "0 to 5 V DC", "1 to 5 V DC" or "0 to 10 V DC". Please consult us.

■ Standard specifications
Display

Rated input

Accuracy(Setting, Indication)

Input sampling period

Control output

FBP resolution
Feedback resistanceControl action

EVT output

PV display: 11-segment LCD 5-digit, backlight Red/Green/Orange, Character size: ACD: 24.0x11.0mm(HxW), ACR: 14.0x5.4mm(HxW)
SV/MV/TIME display : 11-segment LCD 5-digit, backlight Green, Character size: ACD: 14.0x7.0mm (HxW), ACR: 10.0x4.6mm(HxW)
MV/DV bar graph: 22-segment LCD bar graph, backlight Green
MEMO/STEP display : 11-segment LCD 2-digit, backlight Orange, Character size: ACD: 10.0x5.0mm (HxW), ACR: 10.0x4.6mm(HxW)
Thermocouple : K, J, R, S, B, E, T, N, PL-Ⅱ, C(W/Re5-26), External resistance, 100Ω or less(However, B input: External resistance, 40Ω or less)
RTD: Pt100, JPt100, 3-wire system Allowable input lead wire resistance: 10Ω or less per wire
DC current: 0-20mA DC, 4-20mA DC Input impedance: 50Ω Allowable input current, 50mA or less
DC voltage: 0-10mV DC, -10-10mV DC, 0-50mV DC, 0-100mV DC, 0-1V DC:
Input impedance: 1MΩ or more
Allowable input voltage: 5V DC or less
Allowable signal source resistance: 0-10mV DC: 20Ω or less, -10-10mV DC: 40Ω or less, 0-50mV DC: 200Ω or less,
0-100mV DC: 200Ω or less, 0-1V DC: 2kΩ or less
0-5V DC, 1-5V DC, 0-10V DC:
Input impedance: 100kΩ or more Allowable input voltage: 15V DC or less Allowable signal source resistance: 100Ω or less
Thermocouple : Within ±0.2% of each input span±1digit,
However R, S input, 0 to 200℃ (32 to 392°F): Within ±6℃ (12°F)
B input, 0 to 300℃ (0 to 572°F): Accuracy is not guaranteed.
K, J, E, T, N input, less than 0℃ (32°F): Within ±0.4% of input span±1digit
RTD: Within ±0.1% of each input span±1digit
DC current: Within ±0.2% of each input span±1digit
DC voltage: Within ±0.2% of each input span±1digit
Cold junction temperature compensation accuracy: Within ±1℃ at 0 to 50℃
125ms (250ms when EA1/EA2 or EV1/EV2 option is added)
ACD-13A, ACR-13A
Relay contact : 1a 1b, Control capacity; 3A 250V AC (resistive load), 1A 250V AC (inductive load cosφ=0.4), Electrical life, 100,000 cycles
Non-contact voltage (for SSR drive): 12V DC±15% Max. 40mA (short circuit protected)
DC current: 4 to 20mA DC (Resolution 1/12000) Load resistance, Maximum 600Ω
ACD-15A, ACR-15A
Relay contact : 1ax2, Control capacity; 3A 250V AC (resistive load), 1A 250V AC (inductive load cosφ=0.4), Electrical life, 100,000 cycles
1/1000 (corresponds to fully open and fully closed by FBP adjustment) (ACD-15A, ACR-15A)
100Ω to 10kΩPID action (with auto-tuning function), PI, PD action (with Auto/Manual reset function), P action (with Auto/Manual reset function), ON/OFF action
OUT1 proportional band (P): 0 to Input span℃(°F) or 0.0 to 1000.0% (ON/OFF action when set to 0 or 0.0) (Default: 10℃)
OUT1 Integral time (I): 0 to 3600sec (OFF when set to 0) (Default: 200sec)
OUT1 Derivative time (D): 0 to 1800sec (OFF when set to 0) (Default: 50sec)
OUT1 proportional cycle (*1)
: 1 to 120sec (Default: Relay contact; 30sec, Non-contact voltage; 3sec, DC current; Not available)
ARW: 0 to 100% (Default: 50%)
OUT1 ON/OFF action hysteresis: 0.1 to 1000.0℃(°F) or 1 to 10000 (The placement of the decimal point follows the selection) (Default: 1.0℃)
OUT1 high limit, low limit: 0 to 100% (DC current: -5 to 105%) (Not available for ON/OFF action) (Default: OUT1 low limit; 0%,
OUT1 high limit; 100%)
MV high limit, low limit (*2)
: 0 to 100% (Not available for ON/OFF action) (Default: MV low limit; 0%, MV high limit; 100%)
Open output time (*2)
: 0.1 to 1000.0sec (Default: 30.0sec)
Closed output time (*2)
: 0.1 to 1000.0sec (Default: 30.0sec) Output time corresponds to the MV 0 to 100%.
Open/Closed output dead band (*2) : 0 to 100% of the proportional band (Default: 10%)Open/Closed output hysteresis (*2) : 0 to 100% of the proportional band (Default: 1%)(*1): ACD-13A, ACR-13A, (*2): ACD-15A, ACR-15A
EVT1 output
Output: Relay contact 1a, Control capacity: 3A 250V AC (resistive load), 1A 250V AC (inductive load cosφ=0.4), Electrical life: 100,000 cycles
EVT2 output (ACD-13A, ACR-13A)
Output: The same as EVT1 If DR/DS/DA or P option is added, EVT2 output is disabled.

EVT output

Supply voltage
Power consumption
Insulation resistanceDielectric strength
Environment
Case Material/Color
Mounting, Setting
Dimensions, Weight
Attached functions

Accessories included

Accessories sold separately

Alarm action
Alarm types: High limit alarm, Low limit alarm, High/Low limits alarm, High/Low limits independent, High/Low limit range,
High/Low limit range independent, Process high alarm, Process low alarm, High limit alarm with standby,
Low limit alarm with standby, High/Low limits with standby, High/Low limits with standby independent
One type can be selected from 24 types (with status Energized/De-energized) and No event. (Default value: No event)Setting accuracy : Based on the Accuracy and Cold junction temperature compensation accuracy
Action: ON/OFF action
Hysteresis: Thermocouple, RTD input : 0.1 to 1000.0℃ (°F)
DC voltage, current input : 1 to 10000 (The placement of the decimal point follows the selection)
Output: EVT output for which alarm is selected during Event output allocation
Loop break alarm
Setting range: Loop break alarm time: 0 to 200minutes
Loop break alarm span: TC, RTD input; 0 to 150℃(°F), 0.0 to 150.0℃(°F)
DC voltage, current input: 0 to 1500 (The placement of the decimal point follows the selection)
Output: EVT output for which Loop break alarm is selected during Event output allocation.
100 to 240V AC 50/60Hz(Allowable fluctuation range: 85 to 264V AC), 24V AC/DC 50/60Hz(Allowable fluctuation range: 20 to 28V AC/DC)Approx. 18VA
10MΩ or more, at 500V DC
Between power terminal and ground: 1.5kV AC for 1 minuteBetween input terminal and ground: 1.5kV AC for 1 minuteBetween input terminal and power terminal : 1.5kV AC for 1 minute
Ambient temperature: 0 to 50℃ Ambient humidity: 35 to 85%RH (Non-condensing) Conforms to RoHS directive. Drip-proof/Dust-proof (IP66)
Material: Flame-resistant resin, Color: Black
Mounting: Flush Setting: Sheet key input
Dimensions: ACD: 96x96x110mm (WxHxD), ACR: 48x96x110mm (WxHxD) Weight: ACD: Approx. 460g, ACR: Approx. 330g
Sensor correction, Set value lock, Auto/Manual control, Program control function, Set value ramp function, Power failure countermeasure,Self-diagnosis, Automatic cold junction temperature compensation, Burnout (overscale), Input error indication, Indication range / Control
range, Warm-up indication, Console communication, PV color selection, Timer function, Bar graph, PID zone function.
Mounting brackets 1 set, Gasket (Front mounted to the unit) 1 piece
Instruction manual 1 copy, Communication instruction manual 1 copy (when C or C5 option is added)
For the ACR only:
Harness EVT5 : 1 piece [When Event output (A5 option) is added]
Harness W: 1 piece [When Heater burnout alarm (W option) is added] (ACR-13A)
Harness W: 2 pieces [When Heater burnout alarm (W3 option) is added] (ACR-15A)
Harness E: 1 piece [When External setting input (EA1, EA2, EV1, EV2 option) is added]
Harness VT : 1 piece [When Transmission output (TA1, TV1 option) is added]
Harness FBP : 1 piece (ACR-15A)
Terminal cover, Heater burnout alarm (W, W3 option): 20A; CT (CTL-6S), 100A; CT (CTL-12-S36-10L1U), USB communication cable (CMB-001)■ Optional specifications

An Event input comprises events from EVI1 to EVI4. Events selected from Event input allocation will be performed depending on the Input ON(Closed) or OFF (Open) status.
If Set value memory function is selected: 20, 21, 22 and 23 will be allocated to Event input EVI1 to EVI4 respectively, and SV1 to SV15 will be
EVT input [EI]
determined by each value of EVI1 to EVI4. The selected memory number is indicated on the MEMO/STEP display.
Circuit current when Closed: Approx. 16 mA
If this option and Serial communication (C, C5 option) are added together, Event input EVI3 and EVI4 cannot be used.
A3 : EVT1 to EVT3 will be added using a common terminal. Output will be turned ON or OFF depending on the conditions selected from Event
output allocation.
Event output [A3] (*)，
If EVT3 (A3 option) is added, Heating/Cooling control (DR/DS/DA option) or Insulated power output (P option) cannot be added together.
Event output [A5]A5 : EVT4 and EVT5 can be added. Output will be turned ON or OFF depending on the conditions selected from Event output allocation.
Output: Relay contact 1a, Control capacity: 3A 250V AC(resistive load), 1A 250V AC (inductive load cosφ=0.4), Electrical life: 100,000 cycles
Rating: Single-phase 20A, 3-phase 20A, Single-phase 100A, 3-phase100A (Selectable by keypad)
Single-phase: Detects burnout with CT1 input 3-phase: Detects burnout with CT1 and CT2 input
Setting range : 0.0 to 20.0A for Heater rated current 20A [W(20A) W3(20A)] (Off when set to 0.0)
0.0 to 100.0A for Heater rated current 100A [W(100A) W3(100A)] (Off when set to 0.0)
Heater burnout alarm
Setting accuracy : ±5% of the rated current[W, W3] (*)
Action point: Set value
Action: ON/OFF action
Output: Relay contact 1a, Control capacity: 3A 250V AC (resistive load), 1A 250V AC (inductive load cosφ=0.4), Electrical life: 100,000 cycles
Heating control action: The same as Control output (OUT1)
Cooling control action:
OUT2 proportional band
: 0.0 to 10.0 times OUT1 proportional band (ON/OFF action when set to 0.0)
OUT2 integral time, OUT2 derivative time: The same as those of OUT1
OUT2 proportional cycle: 1 to 120sec [Default: DR; 30sec, DS; 3sec, DC current (DA); Not available]
Overlap/Dead band setting range : TC, RTD input: -200.0 to 200.0℃(°F),
DC input: -2000 to 2000 (The placement of the decimal point follows the selection)
OUT2 ON/OFF action hysteresis : TC, RTD input: 0.1 to 1000.0℃(°F) (Default: 1.0℃),
Heating/Cooling control
DC input: 1 to 10000 (The placement of the decimal point follows the selection)
Output [DR, DS, DA]
OUT2 high limit, OUT2 low limit : 0 to 100% (DC current output: -5 to 105%) (Not available for ON/OFF action)(*)(Default: OUT2 low limit; 0%, OUT2 high limit; 100%)
OUT2 action mode
: (1) Air cooling (linear characteristic) (2) Oil cooling (1.5th power of the linear characteristic)(3) Water cooling (2nd power of the linear characteristic) (Default: Air cooling)
Output DR: Relay contact, 1a, Control capacity: 3A 250V AC (resistive load), 1A 250V AC (inductive load cosφ=0.4), Electrical life: 100,000 cycles
DS: Non-contact voltage (for SSR drive) 12V DC±15%, Max. 40mA DC (short circuit protected)
DA: DC current 4 to 20mA DC, Resolution (1/12000), Load resistance: Max. 600Ω
If this option is added: Event output (A3 option) or Insulated power output (P option) cannot be added together, and Event output
EVT2 cannot be used.
This option and Console communication cannot be used together. The following operations can be carried out from the external computer.
(1) Reading and setting of the SV (desired value), PID values and various set values(2) Reading of the PV (process variable) and action status (3) Function change
Communication line
: EIA RS-485 (C5 option), EIA RS-232C (C option)
Communication method: Half-duplex communication
Synchronization method: Start-stop synchronization: 9600, 19200, 38400bps Selectable by keypad (Default: 9600bps)
Serial communication Communication speed
Data bit/Parity: 7, 8/ Even, Odd and No parity (Selectable by keypad) (Default: 7 bits/Even parity)[C, C5]
Stop bit: 1, 2 (Selectable by keypad) (Default: 1)
Communication protocol: Shinko protocol/Modbus ASCII/Modbus RTU (Selectable by keypad) (Default: Shinko protocol)
Number of connectable units : 1 unit to 1 host computer (C), Maximum 31 units to 1 host computer (C5)
Communication error detection : Parity, checksum (Shinko protocol), LRC (MODBUS ASCII), CRC-16 (MODBUS RTU)
Digital external setting: Receives digital set values from Shinko programmable controllers (PC-900, PCD-33A with SVTC option).
If this option and Event input (EI option) are added together, Event input EVI3 and EVI4 cannot be used.
SV adds external analog signal to remote bias value.
Setting signal: DC current: 4 to 20mA (EA1 option), 0 to 20mA (EA2 option)
DC voltage: 0 to 1V (EV1 option), 1 to 5V (EV2 option)
External setting input: EA1, EA2: 50mA DC or less, EV1: 5V DC or less, EV2: 10V DC or less
[EA1, EA2, EV1, EV2] Allowable input
Input impedance: EA1, EA2: 50Ω, EV1, EV2: 100kΩ
Input sampling period : 250ms
Converting the value (PV, SV, MV or DV) to analog signal every 125ms, outputs the value in current or voltage. (Default: PV transmission)
Outputs Transmission output low limit value (4mA DC or 0V DC) if Transmission output high limit and low limit value are the same.
Transmission output
Resolution: 1/12000[TA1, TV1]
Output: TA1: 4 to 20mA DC (load resistance, Max. 500Ω), TV1: 0 to 1V DC (load resistance, Min. 100kΩ)
Output accuracy : Within ±0.3% of Transmission output span
Output voltage : 24±3V DC (when load current is 30mA DC)
Ripple voltage: Within 200mV DC (when load current is 30mA DC)
Insulated power output
Max. load current : 30mA DC[P] (*)
If this option is added: Event output (A3 option) or Heating /Cooling control (DR/DS/DA option) cannot be added together, andEvent output EVT2 cannot be used.

(*): Applicable to the ACD-13A, ACR-13A.

■ External dimensions ( Scale : mm )Gasket11.5□96■ Terminal arrangement
21

EVENT
INPUT

22

100to240V AC

3

3

23

EXT
CONT

4

RS-485

YB(RX)(RS-232C)

24

NC

5

25
NO

OUT1

6 3A
7
EVT3

NO

8
EVT2

OUT2/
P24 EVT2
24V
DC
30mA

NO

9

7

24V AC/DCPOWER
SUPPLY

5V

28
EVT5

2

24V AC/DC

5

NC

7
EVT3

19
TC

B
RTD

30 3A

250V
AC

NO

8

18

B

DC

EVT4 NONOOUT2/
P24 EVT2
24V
DC
30mA

NO

9
EVT1 NO

20

7

9

DI3

13

DI4

14

SG COM

15

YA
(TX)

YB
(RX)
(RS-232C)

16

EVT
5

250V
AC

5V
mA

A

1A
250V
AC

250V
AC

□92 ＋0.8
0

18

1V

4

NO

n×96−3 ＋0.50Lateral close mounting
n: Number of units mounted

17

NO
NO

10 3A

250V
AC

12

CT2

NO

8 3A

11

DI2

250V
AC

EVT1

10 3A

RS-485

CT1

6 3A

EVT2

29

NO

EXT
CONT

OUT1

1V

19

B

DC

・ACR

TC

B
RTD

20

・ACR-15A
TRANSMITOUTPUT

2

DI1

21

EVENT
INPUT

22

100to240V AC

3

3

3
4

A

mA
NO

EVENT
INPUT

100to240V AC

3

17

DI1

2

16

NO
250V
AC

1

POWER
SUPPLY1

POWER
SUPPLY

15

・ACD-15A
GND

2

14
DI4

27

250V
AC

250V
AC

13

POWER
SUPPLY

CT2

8 3A
9

DI3

SG COM

CT1

10 3A

10 3A

12

26

250V
AC

EVT1

EVT1 NO

YA
(TX)

11

DI2

TRANSMITOUTPUT□92 ＋0.8
0TRANSMITOUTPUT

2

・ACD

GND

92 ＋0.8
0

24V AC/DC1

POWER
SUPPLY■ Panel cutout ( Scale : mm )

130

2

98.5
104.5 (When terminal cover is used)

23

EXT
CONT

4

RS­485

5
OPEN

NO

W

6

YB
(RX)
(RS­232C)

24

F.B.P.
O

YA
(TX)

25

GND

11

DI2

12

DI3

13

POWER
SUPPLY2

24V AC/DC

SG COM

3

C

250V
AC

15

9

28

A

mA

5

250V
AC

FBP
NO

DC

EVT4 NO

NO

7 3A

18

8

30 3A

250V
AC

19

B

250V
AC

5

DI3

13

10 3A

250V
AC

n×48−3 ＋0.5
0

14
DI4

45＋0.5
0

17

NO
NO

5V
mA

1A
250V
AC

DC

A

Caution

18

1V
B
B
RTD

Lateral close mounting
n: Number of units mounted

15
16

4
NO

EVT1

20

12

SG COM

O

EVT

9

TC

B
RTD

YA
(TX)

11

DI2

W
C

1V

NO

RS­485

YB(RX)(RS­232C)

6

17
5V

29
NO

10 3A

EXT
CONT

OPEN

16

27

EVT5
EVT1

3

CLOSE

7 3A
8

2

4

NO

CLOSE

DI1
EVENT
INPUT

100to240V AC

14
DI4

26

TRANSMITOUTPUT1

POWER
SUPPLY

92 ＋0.80POWER
SUPPLY

DI1

11.5

48

130

GND

98.5
104.5 (When terminal cover is used)

・ACR-13A

・ACD-13A

Terminal cover(sold separately)

Screw type mounting bracket

96

□91
106.2

・ACD

Terminal cover(sold separately)
・ACR

Screw type mounting bracket

91
106.2

GasketIf lateral close mounting is used for the controller, IP66
specification (Dust-proof/Drip-proof) may be compromised,and all warranties will be invalidated.

19
TC

20

■ Solderless terminal

5.8mm or less

3.2mm

！

SAFETY
PRECAUTIONS

5.8mm or lessUse a solderless terminal with an insulation sleeve in which the M3 screw fits.
The torque should be 0.63N･m.
3.2mm

Caution・This controller does not have a built-in power switch, circuit breaker or fuse.
It is necessary to install them near the controller.
・For a 24V AC/DC power source, do not confuse
polarity when using direct current (DC).

● To ensure safe and correct use, thoroughly read and understand the manual before using this instrument.
● This instrument is intended to be used for industrial machinery, machine tools and measuring equipment. Verify

correct usage after purpose-of-use consultation with our agency or main oﬃce.
(Never use this instrument for medical purposes with which human lives are involved.)● External protection devices such as protection equipment against excessive temperature rise, etc. must be installed,
as malfunction of this product could result in serious damage to the system or injury to personnel. Also proper
periodic maintenance is required.
● This instrument must be used under the conditions and environment described in the manual. Shinko Technos Co.,
Ltd. does not accept liability for any injury, loss of life or damage occurring due to the instrument being used underconditions not otherwise stated in this manual.

Caution with respect toExport Trade Control Ordinance
To avoid this instrument from being used as acomponent in, or as being utilized in the
manufacture of weapons of mass destruction(i.e. military applications, military equipment,etc.), please investigate the end users andthe ﬁnal use of this instrument.
In the case of resale, ensure that this
instrument is not illegally exported.

・This catalog is as of September 2021 and its contents are subject to change without notice.
・The photos in this catalog do not show actual usage.
・If you have any inquiries, please consult us or our agency.

SHINKO TECHNOS CO., LTD.
OVERSEAS DIVISION

Head Office : 2-5-1, Senbahigashi, Minoo, osaka, 562-0035, JapanTel: +81-72-727-6100
Fax: +81-72-727-7006

URL: https://shinko-technos.co.jp/e/E-mail: overseas@shinko-technos.co.jp
202109 E05