SB series

SPEC. SHEET

Current Loop Supply

Model:

SBD-FFeatures• Usable with a Field communicator [Power for 2-wiretransmitter built-in (output impedance: 240 )]• Simple wiring by plug-in socket• Compact size
• 3-port insulation (Input-Output-Power)

Model
SBD－□□-□FInput
01: 4 to 20mA DC
Output
01: 4 to 20mA DC
02: 0 to 20mA DC
03: 0 to 12mA DC
04: 0 to 10mA DC

05: 0 to 1V DC
06: 0 to 5V DC
07: 1 to 5V DC
08: 0 to 10V DC
DC voltage

Power supply
0: 100 to 240V AC
1: 24V DC

Output range0 to 1V DC
0 to 5V DC
1 to 5V DC
0 to 10V DC

How to orderSpecify a model.

(e.g.) SBD-0101-0F

Output specifications

DC current

4 to 20mA DC
0 to 20mA DC
0 to 12mA DC
0 to 10mA DC

Spanadjustment
range
97.5 to 102.5%
97.5 to 102.5%
97.5 to 102.5%
97.5 to 102.5%

Output voltage

When the output range lower limit is zero, (even if zeroadjustment results in a negative value), the output value
will not be negative.

Output range

Zeroadjustment
range
0 to 2.5%
0 to 2.5%-2.5 to 2.5%
0 to 2.5%

Power for 2-wire transmitter

Input specifications
DC current : 4 to 20mA DC
Shunt resistor: 50 built-in

Allowable
loadresistance
600 or less
600 or less
1k or less
1k or less

Allowable
loadresistance
100 or more
500 or more
500 or more
1ｋ or moreZeroadjustment
range-2.5 to 2.5%
0 to 2.5%
0 to 2.5%
0 to 2.5%

Spanadjustment
range
97.5 to 102.5%
97.5 to 102.5%
97.5 to 102.5%
97.5 to 102.5%: 24V DC 3V(When load current is 20mA)
Ripple voltage: Within 200mV DC(When load current is 20mA)
Max load current : 25mA DC
Output impedance: 240(Usable with a Field communicator)

Performance
Accuracy: Within 0.2%
Response time: 1sec (0→90%) (Average 0.5sec)
Temperature coefficient: 0.015%/
Insulation resistance : 10M or more, at 500V DC(Input - Output - Power)Dielectric strength : 2.0kV AC for 1 minute(Input - Output - Power)
Isolation: 3-port isolation (between Input - Output - Power)
Note: Input includes power supply for 2-wire transmission(terminals 1, 2, 3).

SBD-F
General specifications

Case: Flame-resistant resin Color: Light gray
Front panel: Polycarbonate
Spring type plug: Polyamide Color: GreenAdjustment: By front potentiometer
Zero adjustment : 2.5%
Span adjustment: 2.5%
Indication:PWR Indicator (Green):Lights when the power to the instrument isturned on.
Flashes every 0.5 seconds if an error has
occurred in non-volatile IC memory.
ERR Indicator (Red):
Flashes every 0.25 seconds if input is 110% ormore and frashes every 0.5 seconds if input is-10% or less.
Output status selection:
Selects output status Normal/Reverse with DIPswitch. (NO.1 OFF: Normal, ON: Reverse)
Power failure buffer: 30msec.
Self diagnosis: The CPU is monitored by watchdog timer,
and when abnormal status is found on theCPU, the unit is restarted by reset.

Ferrules(Phoenix Contact GMBH & CO.)Insulation sleeve attached: Model
Cross sections
2AI0.25-6BU 0.2 – 0.25mm
2AI0.34-8TQ 0.25 – 0.34mm
2AI0.5-8WH
0.34 – 0.5mm
2AI0.75-8GY 0.5 – 0.75mm
2AI1-8RD
0.75 –1.0mm
2AI1.5-8BK
1.0 – 1.5mm
2AI2.5-8BU
1.5 – 2.5mm
Crimping pliers: CRIMPFOX ZA3, CRIMPFOX UD6

Environmental specification
RoHS directive compliance

Circuit configuration and terminal arrangement

PWR indicator
ERR indicator

Zero adjustment
Span adjustment

Power supply+－Output +－+Input

Power supply for －2-wire transmitter +External dimensions (Scale: mm)
DIN railDip switch

Spring type plug

Installation specifications
Power supply:
100 to 240V AC 50/60Hz
Allowable voltage range : 85 to 264V AC
Power consumption: Approx. .5VA
24V DC
Allowable voltage range : 20 to 28V ACPower consumption: Approx. 3.5W
Ambient temperature: -5 to 55℃ (23 to 131 )Ambient humidity: 35 to 85%RH（non-condensing）
Weight: Approx. 80g
Mounting: DIN rail mounting (Be sure to use End plates for
fixing the unit when the unit ismounted on a DIN rail.)
External dimensions: W17.5×H75×D85mm

SHINKO TECHNOS CO., LTD.

Tel: 81-72-727-6100
URL: http://www.shinko-technos.co.jp/e/ E-mail: overseas@shinko-technos.co.jp

20091130