Infrared Thermal Imaging Camera

High Resolution・Multi-purpose Model

InfReC R450 SeriesHigh resolution, high sensitivity, high precision！

Versatile model which is useful in wide variety ofscenes from structure diagnosis to research ofmicro component.
High performanceat affordable priceDiagnosis of structure (Bridge)

PPM (Motor bearing, etc)

Feature 1

Feature 2

Feature 3【Fine image for wider area】 【Wide measurement range】 【Close-up imagewith high resolution！】
Room temp. to 1500℃
FOV：24°×18°
Changing temperature of object
Pixels：480×360
can be measured in one range
Spatial resolution：0.87mrad
without range change

Building diagnosis

Observation ofheat storage/heat release

960×720 pixels ※SR mode
Micro object can be measured
without optional lensThermal test of device

Composite mode support
ease of viewing

Specification
Feature◼Easy switch
with one button
Basic Performance

Side-by-side

Image Display

Fusion( transparency changeable)

Automatic Movie Recording
Feature Built In(R450Pro)

Measuring Functions

Picture-In-Picture( transparency changeable)■Recording movie connected PC automatically by external trigger
input to R450
■Data recording linked with test equipment and filed facility is available
without configuring I/O system

USB2.08 to 14μm－40 to 1500ﾟCSensitivity (NETD)

0.025ﾟC at 30ﾟC (with S/N improvement)

Accuracy±1ﾟC *1

Frame Rate

40Hz

Detector Pixels

480(H)× 360(V) pixels

Recording Pixels

Standard mode： 480(H) × 360(V) , Super Resolution (SR) mode： 960(H) × 720(V) *2

Field of View

24ﾟ(H) × 18ﾟ(V) (with standard lens)

Spatial Resolution

Standard mode： 0.87mrad , Super Resolution (SR) mode ：0 .58mrad equivalent *3

Focal Distance

10cm to infinity (with standard lens) *4

Focus

Auto/ Manual

Auto Function

Auto Scale, Auto Focus, Full Auto

Color Pallets

7 pallets (Rainbow, Brightness, Hot-white, Hot-black, etc.)

Gradation

256 / 32 / 16 / 8 grade

Visible Camera

CMOS camera 5M pixels

Visible/Thermal Fusion

Side-by-side, Fusion （ transparency changeable）, Picture-In-Picture（ transparency changeable）

Display Functions

1 to 8 times continuous zoom (with display positioning scroll), Grid Overlay, 9 images multi-display(replay mode)

Image Quality Improvement

Denoising, Averaging (with ghost rejection), Edge enhancement

Point Temperature

10 Movable Points, Temperature search: MAX/MIN x 1 each, Delta T

Temperature Display in Assigned Region

MAX, MIN and AVG in BOX (Up to 5 boxes)

Line Profile

Horizontal, Vertical, Horizontal & Vertical

2x Wide Angle
Lens

Field of View(H)×(V)
L=1m
Spatial Resolution

21×16cm

43×32cm

86×64cm

Normal Mode

0.45mm

0.9mm

1.8mm

Super Resolution(SR) mode

0.3mm

0.6mm

1.3mm

Options/Model

Accessory

8.0Hz

N/A

N/A

Emissivity, Environmental/Background, Distance

Emissivity

N/A

Drift Stabilizer

Yes

N/A

SD card, Conforms to SDHC

Data Format
Data Storage
Super Resolution (SR)Still image ： JPEG *5 with temperature data,14bit（with visible image）
Movie ： SVX file （ Dedicated format）
Still image(JPEG *5 with temperature data,14bit）, Recorded with visible image
Yes

Quick Panorama

Horizontal equivalent to 75ﾟ/ Vertical equivalent to 56ﾟ

SD Movie Recording

Max 5HzInterval Recording

3 sec to 60 min interval, with Visible image recorded

External Trigger Recording

Yes

Voice Recording

30sec Recording, replay per a Thermal image

Text Annotation
USB2.0

Annotate up to 128 Characters per a Thermal Image. Characters imported from SD Card

Mass Storage
Movie Data Transfer
External Trigger Input

Max 4Hz

N/A
N/A

Yes
Thermal image: Max：Thermal image: Max：
40Hz、Recorded with8Hz、Recorded withvisible image *6
visible image *6
Automatic recording function by external trigger
input

Thermal image: Max： Thermal image: Max：8Hz、40Hz、Recorded with Recorded with visible image
visible image *6*6
N/A

Video Output

NTSC / PAL Switchable

Alarm Output

Contact Signal. No Voltage

N/A

External Trigger Input

Pulse Signal

N/A

3.5" LCD Monitor (with tilt and brightness adjustment), Color View Finder (with tilt adjustment)
Laser Pointer (Red, class 2, IEC 60825-1 2007/2014), LED Light, Remote Controller

Operating Temperature & Humidity-15°C to 50°C, 90%RH (non-condensing)

Storage Temperature & Humidity-40°C to 70°C, 90%RH (non-condensing)Vibration & Shock29.4m/sec2（3G）、 294m/sec2 （30G）

EMC

Conform to CE regulations (Class A)

Dust & splash proof

Protection class IP54 equivalen

Battery Operation

2.5hours (Typ), Rechargeable Li-Ion battery

AC Power

100V － 220V AC, 50/60Hz

Dimensions

Approx. H121mm×W105mm×D195mm (excluding projection)

WeightApprox. 1.3kg (including Battery Pack)

Standard AccessoriesStandard Software

Wired remote Controller x 1, SD Card x 1, USB Cable x1, Neck Strap x 1, Grip Belt x 1,Rechargeable Li-Ion Battery x 1, Software x 1, Operation Manual x 1, and Carrying Case x 1
InfReC Analyzer NS9500Professional
InfReC Analyzer NS9500Standard *7*1 Only the Range 1 at the environmental temperature from 20 to 30ﾟC （ In other condition, it is ±2ﾟC or ±2%.）*2 Still Image Only
*3 This increased resolution results from detecting characteristic within all frames acquired by the SR process and removing such effects as those caused byhand vibration.
*4 For temperature accuracy : 30cm to infinity*5 SAX format for USA and European countries. JPEG is not available.
*6 Thermal image only when image transfer speed at 40Hz
*7 In order to transfer thermal image movie data by R450 /R450-D, you need to upgrade to “ InfReC Analyzer NS9500 Professional ” ( optional software ).

Power Kit

Options
Lens

40Hz

Multi-point Correction, Emissivity Table
Emissivity Reverse Calculation

Storage Device

OthersNormal Lens

8.0Hz

Alarm Signal Output

Temperature Correction

EnvironmentResistance

2x Telephoto Lens

－40 to 650ﾟC

Alarm Display, Alarm Sound, Color Alarm, Alarm Recording,

Alarm Function

InterfaceLens Type

R450-D

Measuring Range

AuxiliaryField of View and Spatial Resolution are the same magnification withmeasuring distance.

R450

Spectral Range

Display

Measuring Distance and F.O.V.

R450Pro-D

Uncooled Focal Plane Array ( Microbolometer )Storage & Output

R450ProR450ProInfrared Detector

Specification/remarks

2x Telephoto lens, IRL-TX02C-B

12ﾟ(H) x 9ﾟ(V)2x Wide Angle Lens, IRL-WX02C-B

48ﾟ(H) x 36ﾟ(V)52μm Closeup Lens, IRL-C052UC

25mm(H) x 18.7mm(V), Working Distance 56mm

AC adaptor, SDI50-9-UD-P14-110V / 220V

110V or 220V

3-terminal Battery Pack , 2NCR18650BF-G-0077

2800mAh

3-terminal Battery Charger , BC2S1P-110V / 220V

110V or 220V

R450 series is not including AC adapter, battery charger and AC cables.
Please order below Power Kit With R450 series main unit.

■Listed specifications, appearance and design are subject to change without notice. ■Company and commodity names are trade names or registered trade marks of each company. ■NIPPON AVIONICS Co., Ltd. will not be responsible
for any damage of infrared detectors due to incoming strong light (e.g. laser) through lens(es). ■This product is subject to Japanese Export Control Law. Depending on its destination, prior assessment and authorization may be required.
When exporting from country of initial purchase destination, please be sure to follow that country's export regulations as it may require an export permit beforehand.

CAT.NO. 430-023-E 2307-00-Rv5