LINEUP

COOPDECH Syrinjector
COOPDECH Balloonjector

PCA set (Mono flow type)

COOPDECH Syrinjector

Contents: PCA set/Slide clamp/Stand/Carrying bag/Name lavel/Measuring strap(Balloonjector only)/Introduction for use
Quantity(set):10Pump

Syrinjector

Balloonjector

Capacity（mL）

Mono-ﬂow Flow Rate（mL/h）

12 0200300

PCA BolusVolume(mL)

PCA FillingTime(min)Product Number

10

ISJ12-E10-IP1-E1012-EU

1. 0

1. 020

ISJ12-E10-IP1-E2012-EU

2.0

1. 030

ISJ12-E20-IP1-E3012-EU

4.0

1. 020

ISJ12-E40-IP1-E2012-EU

2.0

3.030

CIE20-20-IP3-E3020-EU

5.0

3.030

CIE30-50-IP3-E3030-EU

3.0

3.030

CIE30-30-IP3-E3030-EU

60mL/Premium 120mL (Disposable Infusion Pump)

COOPDECH Balloonjector
200/300 (Disposable Infusion Pump)

Latex Free

PCA set (Flow selector type)Contents: PCA set/Slide clamp/Stand/Carrying bag/Name lavel/Measuring strap(Balloonjector only)/Introduction for use
Quantity(set):10Pump
Balloonjector

Capacity（mL）

Flow-selector Flow Rate (mL/h)
A

B

A＋B

PCA BolusVolume PCA FillingTime(mL)(min)

Product Number200

2.0

4.0

6.0

3.030

CIE20-2040-IP3-E3020-EU300

3.0

5.0

8.0

3.030

CIE30-3050-IP3-E3030-EU

・Be sure to read through the “Instruction for use” before use.
・Please note that specifications and outward appearance may change without notice for enhancement purposes.

Pioneering the future of medical society

Latex Free

[Agency]International Department: 2-6-2, Ayumino, Izumi-city,Osaka, 594-1157, JapanAll products are designed and manufactured by DAIKEN MEDICAL CO., LTD.

COOPDECH’s product information is available at the corporate website.

http://www.daiken-iki.co.jp/2023. 5
Contents of this catalog are as of May 2023.

（2023B）

Made in Japan

COOPDECH Syrinjector

60mL/Premium 120mL (Disposable Infusion Pump)

COOPDECH Balloonjector
200/300 (Disposable Infusion Pump)

Latex FreeSyringe-type infusion pump using atmosphericpressure has achieved a stable flow rate.

▶ COOPDECH Syrinjector infusion pump uses atmospheric pressure to maintain a stableflow rate.
▶ A visible scale allows you to accurately confirm remaining volume.
▶ Improved infusion port cap enables easy opening and closing.
▶ 60mL and 120mL pumps available, they can be used for various purposes.

Latex FreeA large capacity balloon type achieves stableflow characteristics.

▶ Flow rates from 2.0mL/h to 8.0mL/h is available.
▶ Measuring Strap is available to view remaining volume of drug solution.
▶ 200mL and 300mL capacities are available

The set flow rate is achived by using
a vacuum, or negative pressure.
Cylinder A

Inlet Filler Port(with Vent)

Cylinder B

Drug

Outlet Port

Negative Pressure (Vacuum)

Measuring strap[Flow Rate]

Infusion Consistency and Infused Volume(Syrinjector priming speed 2mL/h)

Flow Rate
Infused Volume

2.0120mL
PremiumType60mL
Regular Type
0.0030Time (hour)

60

120 Infused
Volume(mL)
9060300

Cap for the infusion portwith easy opening/closing

Scales are marked in units of5mL for 120mL type and 2mLfor 60mL.

Flow rate diagram of
COOPDECH Balloonjector
Flow rate [mL/h]

Flow Rate(mL/h)

The volume of residual drug
solutions can be measured bysimply hooking Balloonjector main
body onto the measuring strap.

8.0
7.0
6.0
5.0
4.0
3.0
2.0
1.0
0.001020304050
Time (hour)

6070[Test condition]
Saline solution is infused to CIA30-2040(4mL/h)
at room temperature of 23℃

80

COOPDECH Syrinjector
COOPDECH Balloonjector

PCA Device

Pain control can be operated easily.

▶ The device is aimed at reducing the load by PCA device with an easy to press button.
▶ The structure prevents administration of drug solution during the lock-out time.

Light and easy to press infusion button
with a clicking sensation.

Infusion button

Catheter onpatient sideClick

During the lock-out time,the drug solution is not infused,even if the button is pressed. *Valve
Flow regulator

Moderate force is required topress the button that opens
the valve. There is no needfor strong force to press in.

Expansion

The plastic spring structure makesa clicking sound when the button ispressed.

Push reservoir

Reservoir

Easy priming is
Two color-codedenabled.
types of PCA doses.

Catheter onpatient side

Pump side

Catheter onpatient side

Pump side

Catheter onpatient side

The device is designed so that when
the reservoir contracts, the push
reservoir expands, therefore thedrug solution gradually goes fromthe push reservoir through the flow
regulator to the reservoir. Because
the button does not reach the valve
while the drug solution is being
filled, the solution is not infused,even if the button is pressed. Thebutton becomes effective only after
filling is complete.

PressPump side

Lock-out timeExpansion
Catheter onpatient side

The PCA device has two
types of 1mL and 3mL. They
can be used for various
purposes including IV-PCA
and epidural PCA.

The PCA device can be easily
primed by setting it upside-downon the accessory stand.

End of infusion

Integrated PCA set to prevent
misconnection.
The PCA set comes ready to use and requires noassembly. Furthermore, the infusion port isequipped with an anti-needle valve to prevent drug
solutions from extraction by any thin rods such as
injection needles.

Immediately before the end of infusion, thevalve reaches the bottom and is pushed up,
closing the flow channel between thereservoir and catheter on the patient side.

Pump side

Catheter onpatient side

Pump side

Catheter onpatient side

Start of infusion
The valve slides when the button ispressed, the pump and catheter on thepatient side is connected, and the drug
solution in the reservoir is pushed tocatheter on the patient side.

Pressing the infusion button just before the set lock-out time may cause a small infusion of drug solution, resulting ina prolonged lock-out time.
Lock-out time may change depending on factors such as the pump pressure, environmental temperature, type of
medicine and channel.

PARTS

COOPDECH Syrinjector
COOPDECH Balloonjector

Necessary functions are integrated into
each part.
Flow rate can be selected from flow selector type with a flow rate switchover
function and simple mono flow type.
Priming is also simplified by using a priming cap.

LINEUP

COOPDECH Syrinjector
COOPDECH Balloonjector

Pump set (Mono flow type)Contents: Pump set/Carrying bag/Name lavel/Measuring strap(Balloonjector only)/Introduction for use
Quantity(set):10
Pump

Capacity（mL）
60

Syrinjector
12 0Priming cap

Air-vent Filter

Mono flow typePriming cap with a built-inhydrophobic filter is installed at thetip of the line connector. Priming canbe easily done without using tools
such as clamps.

It effectively removes not only the airthat has escaped, but also bubbles
and minute particles that appear in
the drug solution with a rise intemperature.

Mono flow type for single flow rate
employs a spiral tubule with flow rate
precision and reduced blockage.
The pump set comes with a 3-way
stopcock useful for bolus
administration.

Pump

Capacity（mL）
60

200
Balloonjector
300

The flow selector type with a flowrate switchover function features twobuilt-in spiral tubules (flow rate
control tubules), allowing selection of
three types of flow rate to suitdifferent situations.

The pump set employs an IQ valve
with a two-way check valve thatfacilitates the addition and removalof drug solutions. Drug solutions
never reflux during filling, and noliquid leaks out during syringe
removal. The force with which a drug
solution is added has been reduced
as well.

1. 0

I S J 6 -10 - E U

2.0

ISJ6 -20 -EU

1. 0

I S J12 -10 - E U

2.0

I S J12 - 2 0 - E U

3.0

I S J12 - 3 0 - E U

4.0

I S J12 - 4 0 - E U

5.0

I S J12 - 5 0 - E U

Contents: Pump set/Carrying bag/Name lavel/Measuring strap(Balloonjector only)/Introduction for use
Quantity(set):1012 0

IQ Valve

Product Number

Pump set (Flow selector type)

Syrinjector

Flow selector typeMono-ﬂow Flow Rate（mL/h）

Flow-selector Flow Rate (mL/h)
A

B

A＋B

Product Number

0.5

1. 0

1. 5

I S J 6 - 0 510 - E U

1. 0

2.0

3.0

I S J 6 -10 2 0 - E U

4.0

8.0

12 . 0

ISJ6 -4080 -EU

0.5

1. 0

1. 5

I S J12 - P 0 510 - E U

1. 0

2.0

3.0

I S J12 - P10 2 0 - E U

2.0

3.0

5.0

I S J12 - P 2 0 3 0 - E U2.04.0

6.0

I S J12 - P 2 0 4 0 - E U

4.08.012 . 0I S J12 - P4 0 8 0 - E U

2.0

3.0

5.0

CIA 20 -2030 -EU

2.0

4.0

6.0

CIA 20 -2040 -EU

2.0

4.0

6.0

CIA30 -2040 -EU

3.0

5.0

8.0

CIA30 -3050 -EU