DATA SHEET APOLLO MR3T

page 1 of 2

ADVANCED LASER SYSTEM FOR PATIENT ALIGNMENTAPOLLO MR3T red

APOLLO MR3T green

APOLLO MR3T blue

DESCRIPTION
Room lasers are necessary for patient positioning, aligning phantoms forregular quality assurance checks, and for assuring optimized isocenter
accuracy for MR-Linacs.
Integrated in a high-quality aliminium housing, our APOLLO MR3T room
lasers combine advanced optics, high-precision mechanics and stateof-the-art electronics to ensure the highest level of stabiliy and reliability.
The APOLLO MR3T system is certified for MRI devices up to 3 Tesla.

900–1400 mm
Variable Isocenter
HeightINCLUDED■■MR-conditional roomlaser as configured■■Adjustable tilting brackets■■Central power supply■■IR remote control■■Room planning, installation and commissioning

Free-standing column■■12 months standard warranty

OPTIONS■■Free-standing, floor mounted MR conditional column (post) for isocenter

height from 900 to 1400 mm (from 35.4 to 55.1 inch)■■AQUARIUS phantom

Laser Adjustment■■All 6 degrees of freedom canbe adjusted via remote control■■Shift, tilt, rotation■■FocusDATA SHEET APOLLO MR3T

page 2 of 2

TECHNICAL DATASYSTEM
Dimensions (H × W × D)
WeightInternational Protection Rating
Operating temperature
Ambient conditions
Swivel angle

221 × 110 × 100 mm (8.7 x 4.3 x 3.9 inch)
2.6 kg
IP20
15 … 30°C
35 … 90 % rel. humidity, non-condensing
45°

L ASER
Laser colour (typical wave length)
Laser class
Focusable range
Line length at 3 m distanceLine width up to 4 m distanceMax. laser output powerred (638 nm), green (520 nm), blue (450 nm)
21…4m>3m< 0.5 mm (blue), < 1 mm (red, green)< 1 mW

POWER SUPPLY
External power supply
Internal voltage

100 … 240 V AC, 50 … 60 Hz
24 V DCREMOTE CONTROL
163 × 63 × 21 mm (6.4‘‘ x 2.5‘‘ x 0.8‘‘)
130 g
IP40
6 V DC (4 batteries type AAA/LR 03/Micro, 1.5 V)

www.lap-laser.com
LAP GmbH
Laser Applikationen
Zeppelinstrasse 2321337 Lueneburg
Germany
Phone +49 4131 9511-95
Fax+49 4131 9511-96
Email info@lap-laser.com

LAPLAPof America,
GmbH of LAP GmbH Laser Applikationen.
LAP Laser Applications
APOLLO LLC
MR3T is a registeredLAPtrademark
161Further
Commerce
Rd., Suiteof3 products orLaser
Applikationen
designations
services
may be registered trademarks of Asia Pacific Pte. Ltd.
BoyntonBeach, FL33426
МосквеBlk 750A #07-08
LAP GmbH
Laser
Applikationen orПредставительство
other organizations; вtheiruse by third parties
USAmay infringe the rights of the respective
1, Казачийпереулок 7
Chai Chee Roadowners.
Phone +1 561 416-9250
119017 Москва
Technopark@Chai Chee
Fax+1 561 416-9263
Российская Федерация
Singapore 469001
ISO 9001
Email america@lap-laser.com
Тел.
+7 495 7304043
Phone +65 6536 9990
ISO 13485
Факс +7 495 7304044
Fax+65 6533 6697
Email info-russia.med@lap-laser.com
Email info-asia.med@lap-laser.com

LAP Laser Applications
China Co. Ltd.
East Unit , 4F Building # 10
LASER RADI
LuJiaZui Software
Park ATIONDO NO T STARE IN TO BEAM
No. 61 Lane 91 EShan Road
CLASS 2 LASER PRODUCT
Shanghai 200127
Complies with 21 CFR 1040.10 and 1040. 11
for deviations pursuant toChinaLaserexcept
Notice No. 50, dated June 24, 2007.
Phone +86 (21)5047-8881
Fax+86 (21)5047-8887
Email info-asia.med@lap-laser.com© L AP GMBH, MKT-190003 1.0 en, 2019-04-18

Dimensions (H × W × D)
WeightInternational Protection Rating
Power supply