16CHﾊﾟﾙｽﾓｰﾀｺﾝﾄﾛｰﾗ
ＰＭ１６Ｃ－０４ＸＤ(Ｌ)取 扱 説 明 書（ﾘﾓｰﾄﾓｰﾄﾞ操作編）
2658,2735（改訂17/171013）

APPLICATIONOFELECTRONIC

DEVICESツ ジ 電 子 株 式 会 社
本社／〒300-0013 茨 城 県 土 浦 市 神 立 町 3 7 3 9
TEL.029-832-3031㈹ FAX.029-832-2662
URL
http://www.tsujicon.jp/
E-mail info2@tsuji-denshi.co.jp

COMMAND LIST（LAN,RS232C,GP-IB)
x:0～F(motor)

dd･･･d:decimal data

mode
R
R
R
R
R
R/L

command
ABSx±dddddddddd
ABSxB±dddddddddd
ABSxS±dddddddddd
ASSTP,AESTP
BLx±ddddddd
BL?x

R
R
R
R/L

CSCANPx,CSCANNx
FDHPx
FLx±ddddddd
FL?x

R
R
R
R/L
R
R
R/L

GTHPx
JOGPx,JOGNx
LOC
LS?
PAUSE ON, PAUSE OFF, PAUSE?
PSx±ddddddd
PS?x

R
R
R
L
R
R/L
R
R/L
R
R/L
R

RELx±dddddddddd
RELxB±dddddddddd
RELxS±dddddddddd
REM
RTExddd
RTE?x
SABSx±ddddddd
SABS?xSACTxY
SACT?xSCANHPx,SCANHNx

RR

SCANPx,SCANNxSETHPx0XYZ

R/L
R
R/L
R

SETHP?xSETJGxdddd
SETJG?x
SETLSxDYYY0yyy

R/L
R

SETLS?x
SETMTxABCD

reply
absolute index scanabsolute index scan,both dir.backlash rejectabsolute index scan,single dir.backlash reject
all moving motor slow stop or fast stop
set backward digital limit
read backward digital limit
reply:±ddddddd(decimal)
constant speed scan to cw or ccw direction
find home position(start auto find sequence)
set forward digital limit position
read forward digital limit
reply:±ddddddd(decimal)
go to home position if it exists
Jog command(one step to cw or ccw direction)
Local mode change
LS read
reply:abcdHJKL
set pause, clear pause, read out/reply: ON or OFF
preset position dataread position data
reply:±ddddddd(decimal)
relative index scanrelative index scan,both dir.backlash remove
relative index scan,single dir.backlash remove
Remote mode changeacc. rate set
ddd:0～115(see rate table p)
read acc. RATE
reply:ddd
set abs pos data for local mode
read abs pos data for local mode reply:±ddd･･･dd
set active for local Y:0/non active 1/active
read active for local reply:0 or 1
accelerative scan to cw or ccw direction
if HP switch then stop
accelerative scan to cw or ccw direction
HP find information set
X:found/1,not found/0
Y:found dir. 0/cw,1/ccw
Z:auto start dir. 0/cw,1/ccw
read HP find information reply:0XYZ
set Jog pulse for manual PB
read JOG pulse for manual PB
reply:DDDD
se LS characteristics
D:digital limit enable/1, disable/0
Y:LS enable/1, disable/0
y:LS N.C/1, N.O/0
read LS setting
reply:DYYY0yyy(see SETLSx)
motor drive set
A:1/drive enable 0/disable
B:1/hold on
0/hold off
C:0/const
1/trapezoidal 2/S character
D:0/Pulse-Pulse 1/Pulse-Direction
2/PDR

mode
R/L
R
R/L
R
R/L
R
R
R/L
R/L

command
SETMT?x
SHPx±ddddddd
SHP?x
SHPFxdddd
SHPF?xSPCxdddd･･･
SPDHx,SPDMx,SPDLx
SPD?x
SPDAL?

R
R/L
R
R/L
R
R/L
R
R/L

R
R

SPDHxdddd･･･
SPDH?xSPDLxdddd･･･
SPDL?xSPDMxdddd･･･
SPDM?xSPRSx±ddddddd
SPRS?xSRELx±ddddddd
SREL?x
SRQx1(ONLY FOR GP-IB)
SRQx0(ONLY FOR GP-IB)
SRQ?x(ONLY FOR GP-IB)
SSTPx,ESTPx
STOPMDxAB

R/L
R/L

STOPMDx?
STQ?

R/L

STS?

R/L

STSx?

R
R/L
R
R/L
R

TMGEx±ddddddd
TMGE?xTMGIxddddddd
TMGI?xTMGMxY

R/L

TMGM?x

R/L
R/L(available from V1.47)

reply
read motor set
reply:ABCD
set home position dataread home position reply:±ddddddd or NO H.P
set home position offset
read home position offset reply:dddd
change speed while moving(１～5,000,000)
change speed
read speed
reply:HSPD or MSPD or LSPD
active axis and their set speed read
reply:
abcd/Hxxxxxxx/Mxxxxxxx/Lxxxxxxx/Hxxxxxxx(busy axis will reply "000000")
set HSPD to ddd･･･ in pps unit(１～5,000,000)
read HSPD
reply:dddddd
set LSPD to ddd･･･ in pps unit(１～5,000,000)
read LSPD
reply:dddddd
set MSPD to ddd･･･ in pps unit(１～5,000,000)
read MSPD
reply:dddddd
set preset data for local mode
read preset data for local mode reply:±DDD･･･DD
set rel pos data for local mode
read rel pos data for local mode reply:±DDD･･･DD
SRQ flag set
auto reset after SRQ out
SRQ flag reset
SRQ flag read
reply:1 or 0
slow stop or fast stop
set PB and LS stop mode
A:0/LS slow stop 1/LS fast stop
B:0/PB slow stop 1/PB fast stop
read PB & LS stop mode reply:AB
A,B:0,1Remote/Local mode and moving motor information
reply: Rn or Ln n:0 ～ 4 stopping motor number(if n = 0 then you can't start another motor)
Status read out
reply:
R(L)abcd/PNNS/VVVV/HHJJKKLL/±uu･･･/±vv･･･/±ww･･･/±xx･･･
PNNS:P:cw moving N:ccw moving S:stopped
VVVV:LS status & hold off status
HH,JJ,KK,LL:mcc statusuu･･･,vv･･･,ww･･･,xx･･･ :current position
Channel x status read out
reply:R(L)aPVHH±uu･･･
reply data and sequence are matched with
"STS?" command except "/".
timing out end point set
read timing out end point
reply:±DDD･･･DD
timing out interval step set
read timing out interval step reply:DDD･･･DD
timing out mode set
Y:0/disable 1/gate 2/200ns 3/10us 4/100us
5/1ms pulse out
read timing out mode reply: 0 ～ 5

mode
R
R/L
R
R/L
R

R

R

R

R

R

R

R

R

R

R

command
reply
TMGRx
TMGCx
timing out ready set or ready clear
TMGR?xread timing ready
reply: YES or NO
TMGSx±ddddddd
timing out start point set
TMGS?xread timing out start point reply:±DDD･･･DD(Continuous) interpolation function command
A pair of position controller A,B or position controller C,D is available"C0･･･" command is to use controller A,B pair"C1･･･" command is to use controller C,D pair
Speed of synthesis depend on the motor that is seto to A,C controller
C0ALNab±xx･･･/±yy･･･
absolute address a,b axis straight line
C1ALNab±xx･･･/±yy･･･interpolative move to the destination.
a,b:0～F xx･･･,yy･･･:decimal under 9 digit
C0RLNab±xx･･･/±yy･･･
relative address a,b axis straight line
C1RLNab±xx･･･/±yy･･･interpolative move to the destinationa,b:0～F xx･･･,yy･･･:decimal under 9 digit
C0ACPab±xx･･･/±yy･･･/±uu･･･ absolute address a,b axis cw direction circular/±vv･･･
interpolation with center position data.
C1ACPab±xx･･･/±yy･･･/±uu･･･
a,b:0～F x,y:destination u,v:center position/±vv･･･
x･･･,y･･･,u･･･,vv･･･：decimal under 9 digit
perfect circle interpolation will be done bysetting the destination to currnt position.
C0ACNab±xx･･･/±yy･･･/±uu･･･ absolute address a,b axis ccw direction circular/±vv･･･
interpolation with center position data.
C1ACNab±xx･･･/±yy･･･/±uu･･･
another data are same as above.
/±vv･･･
C0RCPab±xx･･･/±yy･･･/±uu･･･ relative address a,b axis cw direction circular/±vv･･･
interpolation with center position data.
C1RCPab±xx･･･/±yy･･･/±uu･･･
a,b:0～F x,y:destination u,v:center position/±vv･･･
x･･･,y･･･,u･･･,vv･･･：decimal under 7 digit
perfect circle interpolation will be done bysetting the destination to (0,0) position.
C0RCNab±xx･･･/±yy･･･/±uu･･･ relative address a,b axis ccw direction circular/±vv･･･
interpolation with center position data.
C1RCNab±xx･･･/±yy･･･/±uu･･･
another data are same as above.
/±vv･･･
C0AACab±xx･･･/±yy･･･/±uu･･･ absolute address a,b axis circular interpolation/±vv･･･
with bypass point data.
C1AACab±xx･･･/±yy･･･/±uu･･･
a,b:0～F x,y:destination u,v:bypass point/±vv･･･
x･･･,y･･･,u･･･,vv･･･：decimal under 9 digit
C0RACab±xx･･･/±yy･･･/±uu･･･ relative address a,b axis circular interpolation/±vv･･･
with bypass point data.
C1RACab±xx･･･/±yy･･･/±uu･･･
a,b:0～F x,y:destination u,v:bypass point/±vv･･･
x･･･,y･･･,u･･･,vv･･･：decimal under 7 digit
C0ACCab±xx･･･/±yy･･･/±uu･･･ absolute address a,b axis perfect circule/±vv･･･
interpolation with two bypass point data.
C1ACCab±xx･･･/±yy･･･/±uu･･･
a,b:0～F x,y:bypass point 1 u,v:bypass point 2/±vv･･
x･･･,y･･･,u･･･,vv･･･：decimal under 9 digit
C0RCCab±xx･･･/±yy･･･/±uu･･･ relative address a,b axis perfect circule/±vv･･･
interpolation with two bypass point data.
C1RCCab±xx･･･/±yy･･･/±uu･･･
a,b:0～F x,y:bypass point 1 u,v:bypass point 2/±vv･･･
x･･･,y･･･,u･･･,vv･･･：decimal under 7 digitNew added command.
V1.13～
mode

command(Auto change scan command)R
ACSxyyy/PNT/JJJ･･J/FNC/LLL･･L

R

ACS?xyyy

R
R/L

ACSPx
ACSP?xR
R

ACSCx
PAUSE ON
PAUSE?
ABSx(B)±dddddddd
RELx(B)±dddddddd
SCANPxSCANNxPAUSE OFF

R

R
V1.14～mode
R/L

HDSTLS?

R

SETCHabcd

R/L

SETCH?

V1.17～mode
R/L

HOLD?xR

HOLDxON
HOLDxOFF

reply
set data command
x :channnel 0,1,2,･･･,9,A,B,C,D,E,F
yyy:sequence number 0 to 127max
PNT:END should be written at next to end seq.
:ADD relative address from start point:TIM relative time from previous point:ACC speed data while accelerration:DEC speed data while deceleration
JJJ:relative address ±2,147,483,647max:time 0 to 65,535ms
FNC:SPD LLL･･･L means speed data in pps:RTE LLL･･･L means rat data
:SLW slow stop
:FST fast stop
:NOP no operation(skip this seq. number)
LLL:speed 1 to 5,000,000pps:rate rate data number in the same range
data read command
x :channnel 0,1,2,･･･,9,A,B,C,D,E,F
yyy:sequence number 0 to 127max
reply: xyyy/PNT/JJJ･･･J/FNC/LLL･･･L
auto change scan ready set command
auto change scan ready read command
reply:ready, not readyauto change scan ready clear command
send before scan start command
reply ON or OFF
auto change scan start command when the axis is
auto change scan readysend after scan start command for synchronousscan start of multi channels

commandreply
hard and soft limit sw read
reply:abcdHHHHSSSS abcd/ch, HHHH/hard SSSS/softset ch a,b,c,d to control window A,B,C,Dposition.
a,b,c,d: 0～9,A,B,C,D,E,F(HEX) "-" not changed.
read channel setting to current control A,B,C,D

command

reply
hold on/off state read x:ch(0,1,･･･9,A,B,C,D,E,F)
reply: ON or OFF
hold on set to ch x
hold off set to ch x

V1.39～mode
R/L

commandR

LCDB 100

R/L

CHCMNT?xR

CHCMNT0 AB1

reply
LCD brightness of back light read. 1～100%
reply: LCD BRIGHT 100
LCD brightness of back light write. 1～100
no reply.
Chanel comment readx:ch(0,1,･･･9,A,B,C,D,E,F)
reply cnannelx comment:a (all chanel data)
Channel No 0 comment AB1 write. no reply.

R

LCD_DISP T

LCD disp is TIMING OUT PORT No.

R

LCD_DISP C

LCD disp is comment of each channel

LCDB?

V1.42～
mode
R/L

x:0～F:Each channel

SRQG0

R/L

SRQ?G

All channels of SRQ flag via GPIB read.
Reply data is HEX(8000) when channel F active.

R/L

RS_SRQx1

R/L

RS_SRQx0

STOPx reply set via RS232C when channel x motor s
topped. Auto reset after STOPx reply out.
STOPx reply reset via RS232C when channel x motor
stopped.

R/L

RS_SRQG0

All channels of RS_SRQ flag via RS232C are reset.

R/L

RS_SRQ?x

SRQ flag of channnel x is read

R/L

RS_SRQ?GR/L

LN_SRQx1

R/L

LN_SRQx0

R/L

LN_SRQG0

All channels of SRQ flag via RS232C read.
Reply data is HEX(8000) when channel F active.
STOPx reply set via ethernet when channel x motor
stopped. Auto reset after STOPx reply out.
STOPx reply reset via ethernet when channel x mot
or stopped.
All channels of LN_SRQ flag via LAN are reset.

R/L

LN_SRQ?x

SRQ flag of channnel x is read. reply:1 or 0

R/L

LN_SRQ?GAll channels of SRQ flag via LAN read.
Reply data is HEX(8000) when channel F active.

V1.43～modeR

YMDabc

R/L

YMD?

R

YAP±ddd...d

R/L

YAP?

R

YBP±ddd...d

R/L

YBP?

commandcommand

G:All channels
reply
All channels of SRQ flag via GPIB are reset.

reply:1 or 0

reply
Oscillation drive mode set command.
a: drive mode
2: 2 axis
3: 3 axis
b: A-D axis synchro mode 0: disable 1: enable
c: B axis jump drive
0: disable J: enable
Oscillation drive mode status read.
Reply: YMDabc XXXXXXXX: STOP/START/PAUSE
A axis drive pulse set command on
oscillation mode. No reply.
A axis drive pulse read on oscillation mode.
Reply: ±ddddddd
B axis drive pulse set command on
oscillation mode. No reply.
B axis drive pulse read on oscillation mode.
Reply: ±ddddddd

mode
R

command
YBJ±ddd...d

R/L

YBJ?

R

YRPddd...d

R/L

YRP?

R

YCP±ddd...d

R/L

YCP?

R

YCRddd...d

R/L

YCR?

R

YDP±ddd...d

R/L

YDP?

R/L

YSTS?

R/L

YCY?

R
R

YSTRT
YPAUS

R/L

YSTOP

V1.48～mode
R/L

HOLDTM?xR

HOLDTMxddd

commandreply
B axis jump pulse set command on oscillation
mode. No reply.
B axis jump pulse read on B axis jump mode.
Reply: ±ddddddd
Oscillation repeat count set command.
No reply.
Oscillation repeat count read.
Reply: ddd
C axis drive pulse set command on oscillation
mode. No reply.
C axis drive pulse read on oscillation mode.
Reply: ±ddddddd
C axis repeat count set command on oscillation
mode. No reply.
C axis repeat count read on oscillation mode.
Reply : ddd
D axis drive pulse set command on oscillation
mode. No reply.
D axis drive pulse read on oscillation mode.
Reply: ±ddddddd
Read current repeat count of oscillation andC axis. Reply: RPaaa CRccc
aaa: oscillation repeat countccc: C axis repeat count
A axis 1 cycle counter read.
Reply: dddd(msec)
Oscillation drive start command.
Oscillation drive suspend command.
If oscillation drive is suspending, releasesuspending status.
Oscillation drive stop command.
All oscillation driving motor are stopped.
reply
Read hold on time before motor drives.
Reply: xxxms
xxx:hold on time (50～500ms)
Preset hold on time before motor drives.
ddd: 50～500(msec)
no reply.

保守用コマンドmode
R/L
ACSK?xcommand

R/L

ACSH?x

R/L
R/L
R
R/L
R/L
R/L

FROM0, FROM1
FROM?
REST
STSM?
VER?
VERH?

reply
reply
aabb,hhhh,hhhh
aa:code(00:END,01:ADD,02:TIM,80:ACC,81:DEC)
bb:sequence number(0 to 7F)
hh･･･hh:data
reply
aabb,hhhh,hhhh
aa:code(00:SPD,01:RTE,02:SLW,03:FST,0F:NOP)
bb:sequence number(0 to 7F)
hh･･･hh:data
FROM selectcurrent FROM read reply:FROM0,FROM1
restart without power off-on
read status port1 of MCC06 aaaa/bbbb/cccc/dddd
read version reply:1.00 06-10-14 PM16C-04X
hardware version read reply:HD-VER0,1,2,5,6

PM16C-04XDは、PM16Cシリーズの従来のコマンドをそのままお使い頂けます。
以下に、それらのコマンドを示します。

OLD COMMAND LISTCOMPATIBLE COMMANDS with PM16C-02N,-02Z,-04,-04S series.
mode
R/L
R/L
R/L

command
S1R
S1L
S10

R

S11□
S12□
S1301
S1302
S1304
S1308
S130F
S1380
S14
S15□
S16□
S20
S20D
S21
S22
S22D
S23
S24
S24D
S25
S26
S26D
S27
S30□□

R/L
R
R/L

R

S31□□

S32XXXXXX□□orS32XXXXXX□□B
S32□±DDDDDDD or
S32□±DDDDDDDBS33XXXXXX□□orS33XXXXXX
□□B

reply
remote mode change
local mode changeR□□□□motor channel No. reply of "A, B, C, D" for -04 typeR□□
motor channel No. reply of "A, B"
for -02 typeset motor channel □ to A
set motor channel □ to B
SRQ set when A motor stopped
SRQ set when B motor stopped
SRQ set when C motor stopped
SRQ set when D motor stopped
SRQ when A,B,C,D motor stopped (combination available)
SRQ cancel (also with GP-IB COMMAND "IFC")SRQ flag read
R□□
(HEX)
set motor channel □ to C
set motor channel □ to D
A position data read
reply:R□□□□□□ (HEX)
A position data read
reply:±DDDDDDD(DEC)
A position status read reply:R□□(HEX)
B position data read
reply:R□□□□□□ (HEX)
B position data read
reply:±DDDDDDD(DEC)
B position status read reply:R□□(HEX)
C position data read
reply:R□□□□□□ (HEX)
C position data read
reply:±DDDDDDD(DEC)
C position status read reply:R□□(HEX)
D position data read
reply:R□□□□□□ (HEX)
D position data read
reply:±DDDDDDD(DEC)
D position status read reply:R□□(HEX)
A position 2 byte command□□：command
08:+jog,
09:-jog
0C:+CSPD scan
0D:-CSPD scan
0E:+scan0F:-scan
16:pause on
17:pause off
18:hold off
19:hold on 1E:+scan & HP stop 1F:-scan & HP stop
40:slow stop 80:EM.stop
B position 2 byte command□□：command
08:+jog,
09:-jog
0C:+CSPD scan
0D:-CSPD scan
0E:+scan0F:-scan
16:pause on
17:pause off
18:hold off
19:hold on 1E:+scan & HP stop 1F:-scan & HP stop
40:slow stop 80:EM.stop
A position 8 byte command
XXXXXX:HEX position data□□:command
10:CSPD REL.IDX 11:CSPD ABS.IDX
12:REL.IDX
13:ABS.IDX
auto backlash correction with "B"
A position DEC index command＊□: A: absolute index scan, R: relative index scan
±DDDDDDD:DEC position data
auto backlash correction with "B"
B position 8 byte command
XXXXXX:HEX position data□□: command
10:CSPD REL.IDX 11:CSPD ABS.IDX
12:REL.IDX
13:ABS.IDX
auto backlash correction with "B"

mode
R

command
S33□±DDDDDDD or
S33□±DDDDDDDB
S3□
S38□□

S39□□

R/L

S3AXXXXXX□□orS3AXXXXXX□□B
S3A□±DDDDDDD or
S3A□±DDDDDDDBS3BXXXXXX□□orS3BXXXXXX□□B
S3B□±DDDDDDD or
S3B□±DDDDDDDB
S4XY
S4XPS
S4XFL
S4XBL

R

S5XY□□R/L

S5XPS±DDDDDDD
S5XFL±DDDDDDD
S5XBL±DDDDDDD
S6

R

BXhhhh
BXH?
BX±DDDD
BX?
FHPA
FHPB
FHPC
FHPD

reply
B position DEC index command□: A: absolute index scan, R: relative index scan
±DDDDDDD:DEC position data
auto backlash correction with "B"
speed change S34:L SPEED S35:M SPEED S36:H SPEED
C position 2 byte command□□：command
08:+jog,
09:-jog
0C:+CSPD scan
0D:-CSPD scan
0E:+scan0F:-scan
16:pause on
17:pause off
18:hold off
19:hold on 1E:+scan & HP stop 1F:-scan & HP stop
40:slow stop 80:EM.stop
D position 2 byte command□□：command
08:+jog,
09:-jog
0C:+CSPD scan
0D:-CSPD scan
0E:+scan0F:-scan
16:pause on
17:pause off
18:hold off
19:hold on 1E:+scan & HP stop 1F:-scan & HP stop
40:slow stop 80:EM.stop
C position 8 byte command
XXXXXX:HEX position data□□: command
10:CSPD REL.IDX 11:CSPD ABS.IDX
12:REL.IDX
13:ABS.IDX
auto backlash correction with "B"
C position DEC index command□: A: absolute index scan, R: relative index scan
±DDDDDDD:DEC position data
auto backlash correction with "B"
D position 8 byte command
XXXXXX:HEX position data□□: command
10:CSPD REL.IDX 11:CSPD ABS.IDX
12:REL.IDX
13:ABS.IDX
auto backlash correction with "B"
D position DEC index command□: A: absolute index scan, R: relative index scan
±DDDDDDD:DEC position data
auto backlash correction with "B"data read command
X: channel No.(0-F) Y: data No.(0-F)
reply:R□□□□□□ (HEX) 3 byte data from read data address
position data read X: channel No.(0-F) PS: position data read
reply:±DDDDDDD (DEC)forward DGLS read
X: channel No.(0-F) FL: forward DGLS read
reply:±DDDDDDD (DEC)backward DGLS read X: channel No.(0-F) BL: backward DGLS read
reply:±DDDDDDD (DEC)data write command
X: channel No. Y: data No. □□: write data (HEX)
position data write X: channel No.(0-F) PS: position data write±DDDDDDD : write data (DEC)forward DGLS write X: channel No.(0-F) FL: forward DGLS read
±DDDDDDD: write data (DEC)backward DGLS write X: channel No.(0-F) BL: backward DGLS read
±DDDDDDD: write data (DEC)status & LS read (a row of data is B,A,D,C for "-02N" compatible）
reply:R□□□□ □：HEX CODE REM/LOC, LIMIT SW
backlash correction data(for CH X) set(h:HEX,D8F1～270F)
backlash correction data(for CH X) read
reply: Rhhhh(HEX)
backlash correction data(for CH X) set 0～±9999（decimal）
backlash correction data(for CH X) read
reply: ±DDDD(DEC)
Find Home Position of channel A
Find Home Position of channel B
Find Home Position of channel C
Find Home Position of channel D

mode
R

R/L

command
RTHPA
RTHPB
RTHPC
RTHPD
DPX
DNX
GFXDDDD
SPHXDDDDD
SPMXDDDDDSPLXDDDDDSPCXDDDDD
G?X
GF?X
HP?X
SPH?X
SPM?X
SPL?X

reply
Return Home Position of channel A
Return Home Position of channel B
Return Home Position of channel C
Return Home Position of channel Dchange start direction of channel X to positivechange start direction of channel X to negative
write offset data
X: channel
D: decimal
write HSPD X: channel DDDDD: PPS unit speed data
write MSPD X: channel DDDDD: PPS unit speed data
write LSPD X: channel DDDDD: PPS unit speed datachange speed while scanning X: channel DDDDD: PPS unit speed dataread home position status X:0 - F(channel) reply RHH :H ASCII hex
read offset data
X: channel
reply RDDDD :d ASCII dec.
read home position X: channel
reply ±DDDDDDD ASCII 7 dec.
read HSPD of channel X
reply RDDDDD
D: ASCII 5 dec.
read MSPD of channel X
reply RDDDDD
D: ASCII 5 dec.
read LSPD of channel X
reply RDDDDD
D: ASCII 5 dec.

目 次
１．概 要 ・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・２
２．通信ラインの毎の準備 ・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・２
１）LAN(Ethernet TCP/IP)通信の設定 ・・・・・・・・・・・・・・・・・・・・・・・・・・・２２）GP-IB通信の設定
・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・３
３）RS232C通信の設定 ・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・３
３．通信コマンド詳細 ・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・４
１）リモート・ローカル変更コマンド ・・・・・・・・・・・・・・・・・・・・・・・・・・・４
２）SRQ関連コマンド
・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・４
３）ステータス読出しコマンド ・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・４
４）モータコントロールコマンド ・・・・・・・・・・・・・・・・・・・・・・・・・・・・・５① JOGコマンド・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・５② SPEED選択コマンド・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・５③ SCANコマンド ・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・６④ 絶対位置指定SCANコマンド（バックラッシュ除去移動含む） ・・・・・・・・・・・・・・６⑤ 相対位置指定SCANコマンド（バックラッシュ除去移動含む） ・・・・・・・・・・・・・・６⑥ ドライブ中速度変更コマンド ・・・・・・・・・・・・・・・・・・・・・・・・・・・・７⑦ SLOW STOP,FAST STOPコマンド
・・・・・・・・・・・・・・・・・・・・・・・・・・・７⑧ PAUSE ON/OFF コマンド・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・８⑨ 定速SCANコマンド ・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・８⑩ 原点検出ドライブコマンド ・・・・・・・・・・・・・・・・・・・・・・・・・・・・・８
５）パラメータ設定・読出しコマンド ・・・・・・・・・・・・・・・・・・・・・・・・・・・９① モータパラメータ設定・読出しコマンド ・・・・・・・・・・・・・・・・・・・・・・・９② 速度設定・読出しコマンド ・・・・・・・・・・・・・・・・・・・・・・・・・・・・１０③ 加減速RATE設定・読出しコマンド ・・・・・・・・・・・・・・・・・・・・・・・・・１１④ 現在位置設定・読出しコマンド ・・・・・・・・・・・・・・・・・・・・・・・・・・１１⑤ ディジタルリミット値設定・読出しコマンド ・・・・・・・・・・・・・・・・・・・・１２⑥ HP関連フラグ設定・読出しコマンド ・・・・・・・・・・・・・・・・・・・・・・・・１２⑦ ローカルモード用データ設定・読出しコマンド ・・・・・・・・・・・・・・・・・・・１３
６）リミットスイッチ関連フラグ設定・読出しコマンド ・・・・・・・・・・・・・・・・・・１４７）2軸補間ドライブコマンド
・・・・・・・・・・・・・・・・・・・・・・・・・・・・・１５８）タイミングパルス出力機能関連コマンド ・・・・・・・・・・・・・・・・・・・・・・・１６９）走行中のオートチェンジドライブ機能 ・・・・・・・・・・・・・・・・・・・・・・・・１８１０）その他のコマンド ・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・２１１１）追加コマンド・機能 ・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・２２
４．ファームウェアのバージョンアップ ・・・・・・・・・・・・・・・・・・・・・・・・・・２８
５．改訂履歴 ・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・２９６．性能・仕様 ・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・３１- 1 -ＰＭ１６Ｃ－０４ＸＤ(Ｌ)リモートモード操作取扱説明書
１．概 要
PM16C-04XD(L)のローカルモード操作取扱いの詳細は別途PM16C-04XD(L)ローカルモード操作
取扱説明書をご覧下さい。本冊子ではRS232C、GP-IB、LAN(Ethernet TCP/IP)を通してのコントロールについて解説しています。
PM16C-04XD(L)はパネル上の押釦と表示器を使った手元コントロールのほかにお手持ちのパソコンから，通信回線を通してのコントロールができます。
従来のPM16C-02Zシリーズ、PM16C-04，PM16C-04Sのコマンドがそのままお使いいただけます。新たに加えられたコマンドは基本的にチャンネルの切換を意識させない体系をとり、直接モータ番号(0～9,A,B,C,D,E,F)を指定できるようにしました。
２．通信ラインの毎の準備
リモートモードによる運転は、REM/LOC SW のLEDが、REMOTE側が点灯している時のみ
コマンドは受け付けられます。情報の問合せコマンドはいつでも受け付けられます。
この表示がLOC のときは、一度 REM/LOC SW を押しなおすか、外部コマンド "REM" でリモートモードにしてから以下の操作を行ってください。
REMOTEﾓｰﾄﾞ運転は、① LAN② GP-IB③ RS232Cの3種類の通信方法によって行えます。
どの通信方法を選ぶかは自由です。環境に合った通信を選んでください。
それぞれ通信方法によってあらかじめ設定しておくべきことがあります。ご使用になる通信方法によって以下の手順で設定を行ってください。
１）LAN(Ethernet TCP/IP)通信の設定
パネル面から簡単にIPアドレスとポート番号が設定できるようになりました。
(Firmware Version 1.09 ～ )先ず、本体の電源を入れます。
REM/LOCボタンによりLOCにします。
SETUPボタンによりセットアップモードに入ります。
CWボタン(XDL型はCW/CCWレバースイッチ)によりセットアップ画面P9(XDL型はP10)を開きます。
LCD表示器のカーソルを変更したい場所に移動させ、テンキーで変更します。
デフォルトではIPアドレスが、"192.168.1.55" ポート番号が"7777"になっています。
IPアドレスはお使いになっているネットに合わせます。ポート番号はそのままでも結構ですが、変更する場合は、10001～10999が推奨されています。
変更が完了したら、電源を切り再度電源を入れると電源オンのシーケンスで新しい設定に変更されます。
設定が完了したらtelnetなどでクライアントからネット接続後 "VER?"コマンドなどで、通信が成立する事を確かめてみてください。
ゲートウェイIPアドレス、サブネットマスクの設定等その他の詳しい接続の設定方法については、
改訂15以前の"PM16C-04XD(L)リモートモード編取扱説明書の「従来の設定方法」"を参考にしてください。取扱説明書は弊社ホームページのバージョンUP情報のページにPDFファイルで掲載されております。 http://www.tsuji-denshi.co.jp/support/ver_3.html
なお、「従来の設定方法」でもIPアドレスおよびポート番号を設定できますが、起動時にはパネルで設定した値が優先されてセットされます。

- 2 -２）GP-IB通信の設定
GP-IBの設定はパネル面からGP-IBアドレスを設定するだけです。
デリミタは以下の内容で固定になっています。
CR+LF + (EOI)EOIは無くてもPM16C-04XD側では受信できます。
送信時は必ず付加されます。
本体のパネル面から次の手順でGP-IBアドレスを設定します。
先ず、本体の電源を入れます。
REM/LOCボタンによりLOCにします。
SETUPボタンによりセットアップモードに入ります。
CWボタン(XDL型はCW/CCWレバースイッチ)によりセットアップ画面P7(XDL型はP9)を開きます。
行き過ぎたらCCWボタンでページを戻すことができます。
LCD表示器のカーソルをGP-IBアドレスに合わせます。
テンキーにより2～31までの設定ができます。不適切な設定は入力数値が点滅して知らせます。
変更が完了したら、SETUPボタンを押してセットアップモードから抜けると設定は完了です。
電源を再投入する必要はありません。
設定が完了したら"VER?"コマンドなどで、通信が成立することを確かめてみてください。
３）RS232C通信の設定
コネクタ接続は下図のように行います。
パネル側コネクタ:DE9P(ケーブル側コネクタ：DE9S)
ピン番号
１
RXD
２
TXD
３
４
GND
５
６
７
８
９

相手方パソコン
１
２
RXD
３
TXD
４
５
GND
６
７
８
９

RS232Ｃ通信の設定はパネル面からBAUD RATEを設定するだけです。
それ以外のパラメータは以下の内容で固定になっています。
DATA BIT :8
STOP BIT :1 NO FLOW CONTROL NO PARITY
デリミタはCR+LFとなっています。
本体のパネル面から次の手順でBAUD RATEを設定します。
先ず、本体の電源を入れます。
REM/LOCボタンによりLOCにします。
SETUPボタンによりセットアップモードに入ります。
CWボタン(XDL型はCW/CCWレバースイッチ)によりセットアップ画面P6(XDL型はP9)を開きます。
行き過ぎたらCCWボタンでページを戻すことができます。
LCD表示器のカーソルでBAUD RATEを指定します。
BAUD RATEをカーソル移動ボタンの中央にあるCHGボタンで必要な値に変更します。
1200,2400,4800,9600,19200,38400の間で変更できます。
変更が完了したら、SETUPボタンを押してセットアップモードから抜けると設定は完了です。
電源を再投入する必要はありません。
設定が完了したら"VER?"コマンドなどで、通信が成立することを確かめてみてください。

- 3 -３．通信コマンド詳細
コマンドのフォーマットは全てアスキーコードで、デリミタ（区切り文字）CR+LF(0DH+0AH) の受信完了を待って解読作業に入ります。デリミタが無いコマンドを受信すると次のコマンドの最初の文字と見なされ、コマンド受信抜けや解析不能が起こります。
以下のコマンド説明ではこの CR+LF(0DH+0AH)が省略されていますが、必ず付加してください。
１）リモート・ローカル変更コマンド
REMOTE/LOCALどちらのMODEでも受け付けます。
全チャンネルのモータが停止している間のみ変更可能です。
LOC

ローカルモードになります。

REM

リモートモードになります。

２）SRQ関連コマンド
下記のコマンドはGPIB,RS232C,LAN通信によって設定でき、
REMOTE/LOCALどちらのMODEでも受け付けます。
・GPIB関連 SRQコマンド
SRQx1
チャンネルxのモータ停止時にSRQ出力するようにセットします。
一度SRQが出ると自動的にクリアされます。
SRQx0
チャンネルxのSRQフラグをクリアします。
SRQG0
設定されている全部のチャンネルのフラグをクリアします。
SRQ?x
チャンネルxのSRQフラグのセット状態を読み出します。
応答は 1または0です。
SRQ?G
全チャンネルのSRQフラグのセット状態を16進数値で読み出します。
E,Fチャンネルが設定されていれば、応答は（C000)です。
セットされたチャンネルのモータが停止すると、PM16CはGPIBラインのSRQ信号をONします。
この時シリアルポールレジスタをリードすると8ビットのデータは次のように割当てられています。
2^6 1がセットされ、2^0：Aドライブ、2＾1:Bドライブ、2＾2:Cドライブ、2＾3:Dドライブ、
従ってA,B,C,Dのドライブに割り当てられた全チャンネルが停止した場合、返信データは"0x4F"となります。
・RS232C通信関連 SRQコマンド
RS_SRQx1
チャンネルxのモータ停止時にRS232Cラインから"STOPx"が出力するようにセットします。
一度SRQが出ると自動的にクリアされます。
RS_SRQx0
チャンネルxのSRQフラグをクリアします。
RS_SRQG0
設定されている全部のチャンネルのフラグをクリアします。
RS_SRQ?x
チャンネルxのSRQフラグのセット状態を読み出します。
応答は 1または0です。
RS_SRQ?G
全チャンネルのSRQフラグのセット状態を16進数値で読み出します。
E,Fチャンネルが設定されていれば 応答は（C000)です。
・LAN通信関連 SRQコマンド
LN_SRQx1
チャンネルxのモータ停止時にLAN通信から"STOPx"が出力するようにセットします。一度SRQが出ると自動的にクリアされます。
LN_SRQx0
チャンネルxのSRQフラグをクリアします。
LN_SRQG0
設定されている全部のチャンネルのフラグをクリアします。
LN_SRQ?x
チャンネルxのSRQフラグのセット状態を読み出します。
応答は 1または0です。
LN_SRQ?G
全チャンネルのSRQフラグのセット状態を16進数値で読み出します。
E,Fチャンネルが設定されていれば 応答は（C000)です。
３）ステータス読出しコマンド- 4 -REMOTE/LOCALどちらのMODEでも受け付けます。
STQ?

REMOTE/LOCAL MODEの情報と停止中のモータ数を読み出します。
応答は REMOTE MODE のとき Rn
n:0 ～ 4
LOCAL MODE のとき Ln
n:0 ～ 4 となります。
同時に動かせるモータは4台ですので、停止のモータ数が0の時は新たにモータを起動することはできず、起動コマンドは無視されます。
新たにモータを起動する前にこのコマンドで起動できるかどうかを判断します。

STS?

REMOTE/LOCALモード、選ばれている4つのモータの詳しい情報がこのコマンドで殆ど読み出せます。
応答は以下の形式になります。
R(L)abcd/PNNS/VVVV/HHJJKKLL/±uu･･･/±vv･･･/±ww･･･/±xx･･･
ここでR or L はREMOTE MODE or LOCAL MODEを表します。
a,b,c,dは選ばれているモータのチャンネルを表し、0～Fの値になります。
PNNS:P:cw 動作中 N:ccw 動作中 S:停止中を表します。
VVVV:LS status と hold off statusを16進数で表します。
各ビットの意味は b3:hold off b2:hp ls b1:ccw ls b0:cw ls
HH,JJ,KK,LL:4つのモータの現在の状態をそれぞれ2桁の16進数で表します。
各ビットの意味は b7:ESEND
急停止コマンドあり
b6:SSEND
減速停止コマンドあり
b5:LSEND
リミット停止受付
b4:COMERR
エラー
b3:ACCN
減速中
b2:ACCP
加速中
b1:DRIVE
パルス出力中b0:BUSY
データ処理中またはドライブ実行中
uu･･･,vv･･･,ww･･･,xx･･･ :4つのモータの現在位置を表します。
±7桁以下は±7桁で返されそれ以上は数値に応じて桁数が変わります。

(V1.47～)
STSx?

チャンネルxのSTS?コマンドと同等の情報が読み出せます。
応答は以下の形式になります。
R(L)aPVHH±uu･･･
応答の内容、順番は"/"を含まないのを除いて"STS?"コマンドと同じですが、LCDに表示されていないチャンネルを読み出した場合V,HHは"-"で表示されます。

４）モータコントロールコマンド① JOGコマンド
REMOTEでのみ受け付けられます。
JOGPx,JOGNx②

チャンネルxのモータを1パルスだけ動かします。 P:CW方向、N:CCW方向

SPEED選択コマンド
REMOTEでのみ受け付けられます。
SPDHx

チャンネルxのモータの速度選択をHSPDにします。

SPDMx

チャンネルxのモータの速度選択をMSPDにします。

SPDLx

チャンネルxのモータの速度選択をLSPDにします。

REMOTE/LOCALどちらのMODEでも受け付けます。
SPD?x

セットされている速度選択を読み取ります。

- 5 -応答は HSPD or MSPD or LSPD になります。
③

SCANコマンド
REMOTEでのみ受け付けられます。
SCANPx,SCANNx④

チャンネルxのモータをCW方向またはCCW方向へ連続SCANさせます。
最大SCAN速度は②のSPEED選択コマンドで選択された速度になります。

絶対位置指定SCANコマンド（バックラッシュ除去移動含む）
REMOTEでのみ受け付けられます。
ABSx±dddddddddd
チャンネルxのモータを絶対位置指定で移動させます。
dd･･･ : -2,147,483,647～+2,147,483,647 (桁数は任意)
ABSxB±dddddddddd
チャンネルxのモータを絶対位置指定で移動させます。
dd･･･ : -2,147,483,647～+2,147,483,647 (桁数は任意)
先ず、別途指定されているバックラッシュ補正位置に移動し、その後低速で最終位置へ移動します。（必ずバックラッシュ補正移動を行います）
バックラッシュ補正位置が、移動範囲を超えた場合、コマンドは無視されます。
このときはエラーフラグが立ちますので"ERR?"コマンドで読み出すと"BAD ABS COMMAND"と返ってきます。
ABSxS±dddddddddd
チャンネルxのモータを絶対位置指定で移動させます。
dd･･･ : -2,147,483,647～+2,147,483,647 (桁数は任意)
バックラッシュ補正方向と目的位置が同じ方向の場合は直接目的地へ移動します。そうでない場合は先ず、別途指定されているバックラッシュ
補正位置に移動し、その後低速で最終位置へ移動します。
（必要に応じたバックラッシュ補正移動を自動的に行います）
バックラッシュ補正位置が、移動範囲を超えた場合、コマンドは無視されます。このときはエラーフラグが立ちますので"ERR?"コマンドで読み出すと"BAD ABS COMMAND"と返ってきます。

⑤

相対位置指定SCANコマンド（バックラッシュ除去移動含む）
REMOTEでのみ受け付けられます。
RELx±dddddddddd
チャンネルxのモータを相対位置指定で移動させます。
dd･･･ : -2,147,483,647～+2,147,483,647 (桁数は任意)
RELxB±dddddddddd
チャンネルxのモータを相対位置指定で移動させます。
dd･･･ : -2,147,483,647～+2,147,483,647 (桁数は任意)
先ず、別途指定されているバックラッシュ補正位置に移動し、その後低速で最終位置へ移動します。（必ずバックラッシュ補正移動を行います）
RELxS±dddddddddd
チャンネルxのモータを相対位置指定で移動させます。
dd･･･ : -2,147,483,647～+2,147,483,647 (桁数は任意)
バックラッシュ補正方向と目的位置が同じ方向の場合は直接目的地へ移動します。そうでない場合は先ず、別途指定されている
バックラッシュ補正位置に移動し、その後低速で最終位置へ移動します。
（必要に応じたバックラッシュ補正移動を自動的に行います）

- 6 -

⑥

ドライブ中速度変更コマンド
REMOTEでのみ受け付けられます。
SPCxdddd･･･⑦

動作中のチャンネルxのモータ速度をpps単位で変更します。
ddd･･･ : 1 ～ 5,000,000

SLOW STOP,FAST STOPコマンド
REMOTE/LOCALどちらのMODEでも受け付けます。
SSTPx

動作中のチャンネルxのモータを減速停止させます。

ESTPx

動作中のチャンネルxのモータを急速停止させます。

ASSTP

全ての動作中のモータを減速停止させます。

AESTP

全ての動作中のモータを急停止させます。

- 7 -その他の移動関連コマンド⑧ PAUSE ON/OFF コマンド
REMOTEでのみ受け付けられます。

⑨⑩

PAUSE ON
(S3016)

PAUSE ON(動作保留)コマンドです。
このコマンド以降のモータ動作コマンドは保留になり、待機状態になります。
PAUSE OFFコマンドで解除されます。
複数のモータ(2～4台)を同時にスタートさせる場合に使われるコマンドです。
既に動作中のモータには影響しません。

PAUSE OFF(S3017)

PAUSE OFF(動作保留解除)コマンドです。
上記保留状態のモータを保留解除し同時スタートさせます。

PAUSE?

PAUSE 状態を問い合わせます。
応答は、ON や OFF となります。

定速SCANコマンド
REMOTEでのみ受け付けられます。
CSCANPx

チャンネルxのモータをCW方向に定速で移動開始させます。

CSCANNx

チャンネルxのモータをCCW方向に定速で移動開始させます。

原点検出ドライブコマンド
REMOTEでのみ受け付けられます。
FDHPx

チャンネルxのモータの自動原点検出を開始します。
まず、あらかじめ設定された方向に移動を開始し、原点センサーを探します。
リミットスイッチまで行くと停止し、反対方向へ移動します。
自動原点検出開始方向に原点センサーを検出すると通過後に減速停止し、低速で原点センサー方向に戻り、再び原点センサーを検出すると停止します。
自動原点検出開始方向と反対方向に原点センサーを検出すると減速停止し、低速で反対方向に移動し、原点を通過すると停止し、もう一度反対方向に低速で移動し再び原点センサーを検出すると停止します。
上記のようにこのコマンドで原点検出した場合は原点検出方向は
必ず検出開始方向と反対になります。
この位置を原点位置として記憶し、原点有りのフラグを立て、原点検出した
方向も記憶します。
これらの情報は "SHP?x" や "SETHP?x"コマンドで読み出せます。
注）このコマンドで原点検出ができなかった場合はCW LSとCCW LS（ディジタルリミット含む）の間を往復し続けます。
このときは、停止コマンドやSTOPボタンで停止します。
ディジタルリミット値を限界位置近傍(±2147483647付近)に、減速停止するのに十分な距離を置かないで設定するとオーバーフローを起こして位置の符号が反転してしまい、正しい動作が行われません。

GTHPx

再度原点検出行う時に使うコマンドです。
をあらかじめ原点が検出されている時、このコマンドにより原点位置近傍にいち早く移動し、そこから低速で原点を検出します。
原点検出方向は記憶されている方向になります。

- 8 -SCANHPx

チャンネルxのモータの原点検出をCW方向に開始します。
原点を検出すると急停止します。ので、脱調しないように遅い速度設定で行ってください。
原点が見つからなければリミットスイッチで停止します。
おおよその原点場所が分かっている場合に短時間で検出することができます。
このコマンドによる検出の後で"GTHPx"により再検出させるとトータル検出時間を短くできます。

SCANHNx

SCANHPxと同様のコマンドで、原点検出開始方向がCCWになります。

５）パラメータ設定・読出しコマンド① モータパラメータ設定・読出しコマンド
設定コマンドはREMOTEでのみ受け付けられます。
読出しコマンドはREMOTE/LOCALどちらでも受け付けられます。
SETMTxABCD

SETMT?xチャンネルxのモータの基本特性を設定します。
A:1/drive enable 0/disable
動かしたくないモータをdisableにしておくと誤操作を禁止できます。
（enableにしておかないとこのモータは動きません）
B:1/hold on
0/hold off
hold offにしておくと外部設置のドライバに対してhold off信号を出力します。この場合動作コマンドを送る前にhold onにしなければなりません。
また、動作終了後には再びhold off信号を送る必要があります。*1)
手動(パネル)操作ではこれらのシーケンスは自動で行われます。
C:0/const
1/trapezoidal 2/S character
モータの移動形式を定速、台形、S字の中から選べます。
D:0/Pulse-Pulse 1/Pulse-Direction 2/Pulse-Direction-Reverse *2)
外部設置のモータドライバへの信号出力にP-P方式、P-D方式、PDR方式 *2)が選べます。
*1) V1.15～ REMOTEモードでも自動HOLD ON/OFFができるようになりました。
*2) V1.30～ PDRが選択できるのは "VERH?"/"HD-VERn" n > 4 の機種のみ。
上記モータ特性の読出しコマンドです。
応答は ABCD の順で返送され "1010" (default値) などとなります。

HOLDxOFF

モータ停止時に外部のドライバに対してhold off信号を出力します。
モータ動作前にhold off信号が自動で解除され、後述のHOLDTM?コマンドで設定した時間経過後、モータが動作します。モータ停止後500msで再びhold off信号が出力されます。

HOLDxON

hold off信号を出力しません。

HOLD?xhold off信号の設定を読み出すコマンドです。
応答は ON もしくは OFF です。
(HOLDコマンドは、SETMTコマンドで設定できるhold offと同じ内容です。)(V1.48～)HOLDTMxddd

HOLDTM?x

hold off信号出力時、モータ動作前に信号が解除されてから動作するまでの時間を設定します。ddd:50～500の10単位で設定 msec. 単位
初期値は80msに設定されています。
上記HOLDTMコマンドで設定するモータ動作までの時間を読み出します。
応答は dddms です。

- 9 -

②

STOPMDxAB

チャンネルxのモータの停止方法を設定します。
A:0/PB slow stop 1/PB fast stop
パネルのSTOPスイッチで減速停止するか急停止するかを設定します。
B:0/LS slow stop 1/LS fast stop
CW,CCW リミットスイッチで減速停止するか、急停止するかを設定します。

STOPMD?x

上記モータ停止方法の読出しコマンドです。
応答は AB の順で返送され "00" (default値) などとなります。

速度設定・読出しコマンド
設定コマンドはREMOTEでのみ受け付けられます。
読出しコマンドはREMOTE/LOCALどちらでも受け付けられます。
SPDHxdddd･･･

チャンネルxのモータHSPDをddd･･･ ppsに設定ます。

SPDH?x

上記設定値を読み出します。
応答は ddd･･･となります。

SPDMxdddd･･･

チャンネルxのモータMSPDをddd･･･ ppsに設定します。

SPDM?x

上記設定値を読み出します。
応答は ddd･･･となります。

SPDLxdddd･･･

チャンネルxのモータLSPDをddd･･･ ppsに設定します。

SPDL?x

上記設定値を読み出します。
応答は ddd･･･となります。

SPDAL?

現在コントロール中の4つのモータの設定された速度を読み出します。
応答は abcd/Hddddddd/Mddddddd/Lddddddd/Hddddddd となります。
a,b,c,d: 0～F
ddd･･･ : pps単位設定速度(1～5,000,000)（モータ動作中は"0000000"が返されます。

- 10 -RATE DATA TABLE(unit: ms/1000pps)
No.
0
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18
19
③

RATE
1000
910
820
750
680
620
560
510
470
430
390
360
330
300
270
240
220
200
180
160

RATE
150
130
120
110
100
91
82
75
68
62
56
51
47
43
39
36
33
30
27
24

No.
40
41
42
43
44
45
46
47
48
49
50
51
52
53
54
55
56
57
58
59

RATE
22
20
18
16
15
13
12
11
10
9.1
8.2
7.5
6.8
6.2
5.6
5.1
4.7
4.3
3.9
3.6

No.
60
61
62
63
64
65
66
67
68
69
70
71
72
73
74
75
76
77
78
79

RATE
3.3
3.0
2.7
2.4
2.2
2.0
1.8
1.6
1.5
1.3
1.2
1.1
1.0
0.91
0.82
0.75
0.68
0.62
0.56
0.51

No.
80
81
82
83
84
85
86
87
88
89
90
91
92
93
94
95
96
97
98
99

RATE
0.47
0.43
0.39
0.36
0.33
0.30
0.27
0.24
0.22
0.20
0.18
0.16
0.15
0.13
0.12
0.11
0.10
0.091
0.082
0.075

No.
100
101
102
103
104
105
106
107
108
109
110
111
112
113
114
115

RATE
0.068
0.062
0.056
0.051
0.047
0.043
0.039
0.036
0.033
0.030
0.027
0.024
0.022
0.020
0.018
0.016

加減速RATE設定・読出しコマンド
設定コマンドはREMOTEでのみ受け付けられます。
読出しコマンドはREMOTE/LOCALどちらでも受け付けられます。
RTExddd
RTE?x

④

No.
20
21
22
23
24
25
26
27
28
29
30
31
32
33
34
35
36
37
38
39

チャンネルxのモータの加速・減速レートをコードで設定します。
ddd:0～115(RATE DATA TABLE参照)
上記レートの読出しです。
応答 ddd

現在位置設定・読出しコマンド
設定コマンドはREMOTEでのみ受け付けられます。
読出しコマンドはREMOTE/LOCALどちらでも受け付けられます。
PSx±ddddddd

チャンネルxのモータの現在位置を設定します。
ddd･･･ : -2,147,483,647～+2,147,483,647 (桁数は任意)

PS?xチャンネルxのモータの現在位置を読み出します。
応答は ±ddddddd(decimal) となります。
7桁を超えたときはその分桁数が増えます。

- 11 -

⑤⑥

ディジタルリミット値設定・読出しコマンド
設定コマンドはREMOTEでのみ受け付けられます。
読出しコマンドはREMOTE/LOCALどちらでも受け付けられます。
FLx±ddddddd

チャンネルxのモータのCW側のディジタルリミット値を設定します。
ディジタルリミットを有効にしておくとこの値以上のところへは移動しません。
（リミットスイッチが減速停止になっていると、減速停止の期間分
オーバーランします）ディジタルリミットを有効にして現在位置がディジタル
リミット値の範囲外になるような設定をするとモータは範囲内へ向かう
方向のみ動かせます。

FL?x

チャンネルxのモータのCW側ディジタルリミット値を読み出します。
応答は ±ddddddd(decimal) となります。
7桁を超えたときはその分桁数が増えます。

BLx±ddddddd

チャンネルxのモータのCCW側のディジタルリミット値を設定します。
ディジタルリミットを有効にしておくとこの値以下のところへは移動しません。
動作については"FLx"コマンドと同様です。

BL?x

チャンネルxのモータのCCW側ディジタルリミット値を読み出します。
応答は ±ddddddd(decimal) となります。
7桁を超えたときはその分桁数が増えます。

HP関連フラグ設定・読出しコマンド
設定コマンドはREMOTEでのみ受け付けられます。
読出しコマンドはREMOTE/LOCALどちらでも受け付けられます。
SETHPx0XYZ

チャンネルxのモータの原点位置設定方法や状況の強制書き換えを行います。
X:found/1,not found/0
原点検出有り(読み書き可）
Y:found dir. 0/cw,1/ccw
原点検出方向(読み書き可）
Z:auto start dir. 0/cw,1/ccw 自動原点検出開始方向の設定です。

SETHP?xチャンネルxのモータの原点位置検出状態の読出しです。
上記の内容で読み出せます。
応答 :0XYZ の形式で "0100" などと読み出せます。

SHPx±ddddddd

チャンネルxのモータの原点位置データをセットできます。
通常は原点位置データは自動検出されますが、このコマンドにより強制書き換えすることができます。

SHP?x

原点位置データを読み出します。
応答は ±ddddddd(decimal) となります。
7桁を超えたときはその分桁数が増えます。
原点が検出されていないと(検出フラグが立っていないと）返されます。

"NO H.P"がSHPFxdddd

チャンネルxのモータの原点オフセット値を設定します。
GTHPxコマンドで原点の再検出を行うときにこのオフセット値分離れたところか
ら検出を始めます。原点オフセット値の2倍の量の範囲に原点が見つからなければ検出失敗として、原点検出記憶を消します。
dddd:0 ～ 9999 の範囲での設定ができます。
リモートから範囲を超えた設定をしようとすると9999になります。

SHPF?x

チャンネルxのモータの原点オフセット値を読み出します。
応答 dddd 10進4桁で読み出せます。

- 12 -

⑦

ローカルモード用データ設定・読出しコマンド
設定コマンドはREMOTEでのみ受け付けられます。
読出しコマンドはREMOTE/LOCALどちらでも受け付けられます。
この項のデータ値はLOCAL MODE でのみ使われます。
SABSx±ddddddd

LOCAL MODEで使われるチャンネルxの絶対位置移動先の設定です。
LOCAL MODEでABS SCANを選ぶとLCDの下段(XDL型では第二画面)に表示されます。
パネル面からも設定できます。

SABS?xLOCAL MODEで使われるチャンネルxの絶対位置移動先の読出しです。
応答は ±ddd･･･ 10進7桁で読み出せます。
7桁以上はその分桁数が増えます。

SRELx±ddddddd

LOCAL MODEで使われるチャンネルxの相対位置移動先の設定です。
LOCAL MODEでREL SCANを選ぶとLCDの下段(XDL型では第二画面)に表示されます。
パネル面からも設定できます。

SREL?xLOCAL MODEで使われるチャンネルxの相対位置移動先の読出しです。
応答は ±ddd･･･ 10進7桁で読み出せます。
7桁以上はその分桁数が増えます。

SPRSx±ddddddd

LOCAL MODEで使われるチャンネルxのプリセット位置データの設定です。
LOCAL MODEでSCANを選ぶとLCDの下段(XDL型では第二画面)に表示され、PRESETボタンで現在位置をプリセット位置データにすることができます。
パネル面からも設定できます。

SPRS?xLOCAL MODEで使われるチャンネルxのプリセット位置データの読出しです。
応答は ±ddd･･･ 10進7桁で読み出せます。
7桁以上はその分桁数が増えます。

SETJGxdddd

LOCAL MODEで使われるチャンネルxのモータのJOGステップ数の設定です。
LOCAL MODEでJOGボタンによりJOGステップをするとこの値だけ一度に加減速移動で動きます。
dddd:0 ～ 9999 の範囲での設定を推奨します。
パネル面からもセットアップモードに入って設定できます。

SETJG?x

LOCAL MODEで使われるチャンネルxのモータのJOGステップ数の読出しです。
応答は ddd･･･ 10進4桁で読み出せます。
4桁以上はその分桁数が増えます。

SACTxYNo use command.

SACT?xNo use command.

- 13 -

６）リミットスイッチ関連フラグ設定・読出しコマンド
設定コマンドはREMOTEでのみ受け付けられます。
読出しコマンドはREMOTE/LOCALどちらでも受け付けられます。
SETLSxDYYY0yyy

チャンネルxのモータのリミットスイッチの特性を設定します。
D:digital limit enable/1, disable/0
Y:LS enable/1, disable/0 (順にHP LS, CCW LS,CW LS)
y:LS N.C/1, N.O/0(順にHP LS, CCW LS,CW LS)

SETLS?xチャンネルxのモータのリミットスイッチの特性を読出します。
応答は DYYY0yyy となります。(意味は上記参照）

LS?

現在コントロール中のモータ(4台）のチャンネル情報とリミットスイッチの情報を読出します。
応答は abcdHJKL となります。
a,b,c,d:チャンネルの情報
H,J,K,L:リミットスイッチの状態をチャンネルの順に示します。
a chのLS状態はHで、b chはJで、c chはKで、d chはLで示されます。
それぞれ4ビットが16進数で表され
b0:CW LS, b1:CCW LS, b2:HP LS ,b3:hold off となっています。
CW LSのみONの時は1です。CCW LSのみ ONのときは2です。
HP LSのみONの時は4です。hold off出力があると8のビットが立っています。

HDSTLS?

現在コントロール中のモータ(4台）のチャンネル情報とハードリミット、ソフトリミットの情報を個別に読出します。
応答は abcdHHHHSSSS となります。
a,b,c,d:チャンネルの情報
H,H,H,H:HARDリミットスイッチの状態をチャンネルの順に示します。
S,S,S,S:SOFT リミットスイッチの状態をチャンネルの順に示します。
それぞれ4ビットが16進数で表され
b0:CW LS, b1:CCW LS, b2:HP LS(HARDのみ)
CW LSのみONの時は1です CCW LSのみ ONのときは2です。
HP LSのみONの時は4です(HARDのみ)- 14 -

７）2軸補間ドライブコマンド
補間ドライブとは2つの軸をX,Y座標とした平面上で直線、円弧、真円移動をさせる機能です。
これらのコマンドはREMOTEでのみ受け付けられます。
PM16C-04XDは4つの独立したコントローラA,B,C,Dが内蔵されています。
2軸補間動作はコントローラA,Bのペア、C,Dのペアでのみ行われます。
A,Bのペアを使った補間ドライブのコマンドは "C0･･･"で始まります。
C,Dのペアを使った補間ドライブのコマンドは "C1･･･"で始まります。
補間ﾄﾞﾗｲﾌﾞの速度はA,Cｺﾝﾄﾛｰﾗにｾｯﾄされた軸(X軸側)の設定速度になります。
コマンド表
コマンド
応 答 内 容C0ALNab±xx･･･/±yy･･･
absolute address a,b axis straight line
C1ALNab±xx･･･/±yy･･･interpolative move to the destination.
a,b:0～F xx･･･,yy･･･:decimal under 9 digit
C0RLNab±xx･･･/±yy･･･
relative address a,b axis straight line
C1RLNab±xx･･･/±yy･･･interpolative move to the destinationa,b:0～F xx･･･,yy･･･:decimal under 9 digit
C0ACPab±xx･･･/±yy･･･/±uu･･･ absolute address a,b axis cw direction circular/±vv･･･
interpolation with center position data.
C1ACPab±xx･･･/±yy･･･/±uu･･･
a,b:0～F x,y:destination u,v:center position/±vv･･･
x･･･,y･･･,u･･･,vv･･･：decimal under 9 digit
perfect circle interpolation will be done bysetting the destination to currnt position.
C0ACNab±xx･･･/±yy･･･/±uu･･･ absolute address a,b axis ccw direction circular/±vv･･･
interpolation with center position data.
C1ACNab±xx･･･/±yy･･･/±uu･･･
another data are same as above.
/±vv･･･
C0RCPab±xx･･･/±yy･･･/±uu･･･ relative address a,b axis cw direction circular/±vv･･･
interpolation with center position data.
C1RCPab±xx･･･/±yy･･･/±uu･･･
a,b:0～F x,y:destination u,v:center position/±vv･･･
x･･･,y･･･,u･･･,vv･･･：decimal under 7 digit
perfect circle interpolation will be done bysetting the destination to (0,0) position.
C0RCNab±xx･･･/±yy･･･/±uu･･･ relative address a,b axis ccw direction circular/±vv･･･
interpolation with center position data.
C1RCNab±xx･･･/±yy･･･/±uu･･･
another data are same as above.
/±vv･･･
C0AACab±xx･･･/±yy･･･/±uu･･･ absolute address a,b axis circular interpolation/±vv･･･
with bypass point data.(Destination setting to
C1AACab±xx･･･/±yy･･･/±uu･･･ current position aren't allowed, use "C0ACC"/±vv･･･
command) a,b:0～F x,y:destination u,v:bypass poi
nt x･･･,y･･･,u･･･,vv･･･：decimal under 9 digit
C0RACab±xx･･･/±yy･･･/±uu･･･ relative address a,b axis circular interpolation/±vv･･･
with bypass point data.(Destination setting to
C1RACab±xx･･･/±yy･･･/±uu･･･ current position aren't allowed, use "C0RCC"/±vv･･･
command) a,b:0～F x,y:destination u,v:bypass po
int x･･･,y･･･,u･･･,vv･･･：decimal under 7 digit
C0ACCab±xx･･･/±yy･･･/±uu･･･ absolute address a,b axis perfect circule/±vv･･･
interpolation with two bypass point data.
C1ACCab±xx･･･/±yy･･･/±uu･･･
a,b:0～F x,y:bypass point 1 u,v:bypass point 2/±vv･･
x･･･,y･･･,u･･･,vv･･･：decimal under 9 digit
C0RCCab±xx･･･/±yy･･･/±uu･･･ relative address a,b axis perfect circule/±vv･･･
interpolation with two bypass point data.
C1RCCab±xx･･･/±yy･･･/±uu･･･
a,b:0～F x,y:bypass point 1 u,v:bypass point 2/±vv･･･
x･･･,y･･･,u･･･,vv･･･：decimal under 7 digit- 15 -

８）タイミングパルス出力機能関連コマンド
あらかじめ決められたパルスカウンターのスタート点からストップ点の範囲で、ゲート信号(5V TTL)を出力することができます。または、その範囲で決められたパルス毎にタイミングパルス(5V TTL)を発生できます。パルス幅は200ns,10us,100us,1msが選べます。
これらの定数はチャンネル毎に設定できます。
設定コマンドはREMOTEでのみ受け付けられます。
読出しコマンドはREMOTE/LOCALどちらでも受け付けられます。
TMGMxYチャンネルxのモータのタイミングアウトモードの設定をします。
Y:0/disable 1/gate 2/200ns 3/10us 4/100us 5/1ms pulse out
0/disable ではタイミングパルス出力をしません
1/gateではスタート点から終了点の間、TTL(正論理)のゲートパルスを出力します。
2～5ではスタート点から終了点の間、設定されたインターバル毎にそれぞれ
200ns, 10μs, 100μs, 1msのTTL パルス出力をします。

TMGM?x

チャンネルxのモータのタイミングアウトモードの設定を読出します。
応答は 0 ～ 5 となります。

TMGRx

チャンネルxのモータのタイミングアウトをready状態とします。
対象のモータが停止しているときに受け付けられます。
コマンドはパルス出力(タイミングアウトモード 2,3,4,5)のときには、出力開始点より手前にいるときにのみ正しい設定になります。
ready状態とは所定のタイミング出力が出せる準備ができたことを意味します。
ready状態のときモータが終了点に向かってスタート点を通過すると所定の出力が得られます。
パルス出力(タイミングアウトモード2～5)のときは、一度モータがスタート点を通過するとready状態は解除されますが、終了点までパルス出力は行われます。
途中停止した場合は、ready状態が解除されていても、まだ出力されていない点を通過するとパルスは出力されます。
一度パルス出力された点に戻ってもパルス出力はされません。
途中停止して以降パルス出力をしないようにするには、"TMGCx"コマンドを発行
してready解除をしてください。
ゲート出力（タイミングアウトモード1)のときはスタート点を通過してもready状態は解除されません。設定範囲では移動方向にかかわらずゲート信号が出力されます。ready解除してゲート信号を出力しないようにするには"TMGCx"コマンドを発行してreadyを解除します。

TMGCx

チャンネルxのモータのタイミングアウトをready解除します。
モータが停止した状態でのみ受け付けられます。

TMGR?x

チャンネルxのモータのタイミングアウトのready状態を読出します。
応答は ready状態の時 "YES" そうでなければ "NO"となります。
パネル面ではチャンネル表示がnon active（ローカルコントロールができない状態） ＿CH05＿
active(ローカルコントロールができる状態）＊CH05＊
active and timing pulse ready＃CH05＃のように表示されますので、タイミングアウトreadyになっているかどうかの確認ができます。

- 16 -

XDL型では、active状態はLEDランプで timing pulse ready状態は#で示されますTMGSx±ddddddd

チャンネルxのモータのタイミングアウトのスタート点を設定します。

TMGS?x

チャンネルxのモータのタイミングアウトのスタート点を読出します。

TMGEx±ddddddd

チャンネルxのモータのタイミングアウトの終了点を設定します。

TMGE?x

チャンネルxのモータのタイミングアウトの終了点を読出します。

TMGIxddddddd

チャンネルxのモータのタイミングアウトのインターバルを設定します。

TMGI?x

チャンネルxのモータのタイミングアウトのインターバルを読出します。

タイミングアウトの手順は、以下のようになります。
①タイミングアウトを得るコネクタ(A - DまたはTP0-TP3:表示チャンネルA - D に対応)に合わせてモータを内部コントローラにセットします。このとき、チャンネル選択コマンド "SETCHabcd"（詳細は追加コマンド・機能の項参照）
を使います。セットできているかどうかは、"SETCH?"で確認できます。
古いコマンドを使って、以下のように行うこともできます。
A,B,C,Dにセットするコマンドはそれぞれ"S11x", "S12x", "S15x", "S16x"となります。
セットできているかどうかは、"S10"コマンドや"STS?", "LS?"コマンドで確認できます。
（S11x, S12x, S15x, S16x, S10コマンドの詳細はPM16C-04の取扱説明書を参照してください。
本PM16C-04XDは、PM16C-02Z, PM16C-04,PM16C-04Sのコマンドがそのまま使えます）②タイミングアウトを得るチャンネルのモータにタイミングアウトモードの設定("TMGMxY")
デフォルト値: 5/1ms幅パルス
タイミングアウトスタート点の設定("TMGSx±dddddd) デフォルト値: +1,000
タイミングアウト終了点の設定("TMGEx±dddddd)
デフォルト値: +10,000
タイミングインターバルの設定("TMGIxddddddd)
デフォルト値:200を行います。
このデータは一度設定しておくと変更するまで変わりません。またいつ設定してもOKです。
出荷時はデフォルト値が入っています。
③タイミングアウトを得るモータをタイミングスタート点の外側に移動します。
既にその位置になっていればこの操作は省略できます。
ゲート信号出力(タイミングアウトモード1）の場合はこの操作は必要ありませんが、現在位置がスタート点とストップ点の間にあった場合は次項④のready状態にする
コマンドですぐに出力されます。
④タイミングアウト準備完了(ready)状態にします。("TMGRx")
ゲート信号出力(タイミングアウトモード1)の設定がされている場合、現在位置の条件が範囲内にあればすぐに出力されます。
⑤タイミングアウトを得るモータを終了点方向に移動開始します。
絶対位置移動、相対位置移動、連続移動などどれでもかまいません。
これで所定の条件でタイミングパルスが出力されます。
複数の(2～4)チャンネルに異なった条件で設定を行い、同時スタートさせることにより4種類のパルスのタイミング発生器として使うことも可能です。
注）
パルス出力(タイミングアウトモード 2,3,4,5)に設定されていても パルス幅>=パルス出力イ
ンターバル のときには、出力はパルスにならず "H"状態出力になります。

- 17 -

９）走行中のオートチェンジドライブ機能
直線加減速(台形駆動)ドライブ中に速度変更、加減速レート変更などをあらかじめ予約されたタイミングで実行する機能です。S字加減速ドライブに設定していても、直線加減速（台形駆動）になります。
変更タイミングと変更内容は、各軸毎に最大128ヶ所に設定できます。
変更タイミングはスタート点からの相対位置、前の変更タイミングからの時間(ms)、またはドライブパルス速度(PPS)で指定できます。
変更内容は、速度の変更、加減速レートの変更(*1)、停止指示などがあります。
１．データライトコマンド
リモートモードで使用できます。
予約データを書き込むコマンドです。
複数軸の同期スタートを行うときは軸毎に書き込みが必要です。
書換えの必要がなければ再書き込みは不要です。
コマンド形式ACSxyyy/PNT/JJJ･･･J/FNC/LLL･･･L
ACS:Auto Change Speed command(fixed)
x: channel(0 - F)yyy: data number(0 - 127)
PNT:END(no more data)
最終変更点の次は必ずACSxyyy/END/を書き込んでください。
ADD(relative address from start point)
TIM(relative time from previous point)
ACC(speed data while acceleration)
DEC(speed data while deceleration)"JJJ･･･J:相対アドレス(±2,147,483,647)、時間(0-65,535ms)、""
パルス速度データ(1 - 5,000,000pps)"
FNC:SPD(speed in pps)
LLL･･･Lが速度データであることを示します。
RTE(rate number)
LLL･･･Lがレートデータであることを示します。
SLW(slow stop)
SLOW STOPします。(LLL･･･Lは不要です）
FST(fast stop)
FAST STOPします。(LLL･･･Lは不要です）
NOP(no operation)
不要の行を無効にするときに使用します。
"LLL･･･L:パルス速度(1 - 5,000,000pps)、RATE(設定範囲のRATE DATA No.)"(*1)
加減速レートは、現在の設定値の属するグループ内でのみ可能です。
同一グループ内の変更範囲は下の表のとおりです。
現在の属するグループは既に設定してあるRATE No.の属する最上位のグループになります。従って、たとえばM2 TYPEのグループ内でRATE変更を行いたい場合は、
あらかじめRATE Noを85 - 91の範囲に設定しておかなければなりません。
ｸﾞﾙｰﾌﾟ
L1 TYPE
L2 TYPE
M1 TYPE
M2 TYPE
H1 TYPE
H2 TYPERATE No.
0 - 6017 - 76
24 - 84
31 - 91
41 - 100
55 - 115

CH0とCH1を同期ｽﾀｰﾄさせて同じ位置で速度変更する場合のｺﾏﾝﾄﾞ例）
ACS0000/ADD/5000/SPD/3000:ｽﾀｰﾄ点から5000のところで3000PPSへ速度変更
ACS0001/ADD/10000/SPD/6000:ｽﾀｰﾄ点から10000のところで6000PPSへ速度変更
ACS0002/ADD/20000/SPD/10000:ｽﾀｰﾄ点から20000のところで10000PPSへ速度変更
ACS0003/ADD/30000/SLW:ｽﾀｰﾄ点から30000のところで減速停止へ
ACS0004/END:ﾌﾟﾛｸﾞﾗﾑの終了(必ず書き込む）- 18 -

ACS1000/ADD/5000/SPD/3000
ACS1001/ADD/10000/SPD/6000
ACS1002/ADD/20000/SPD/10000
ACS1003/ADD/30000/SLW
ACS1004/END:ｽﾀｰﾄ点から5000のところで3000PPSへ速度変更:ｽﾀｰﾄ点から10000のところで6000PPSへ速度変更:ｽﾀｰﾄ点から20000のところで10000PPSへ速度変更:ｽﾀｰﾄ点から30000のところで減速停止へ:ﾌﾟﾛｸﾞﾗﾑの終了(必ず書き込む）

CH2とCH3を同期ｽﾀｰﾄさせて同じﾀｲﾐﾝｸﾞで速度変更する場合のｺﾏﾝﾄﾞ例）
ACS2000/TIM/0/SPD/1000 (ﾀﾞﾐｰ) :ｽﾀｰﾄ点から0msのところで1000PPSへ速度変更
ACS2001/TIM/1000/SPD/3000:前変更点から1000msのところで6000PPSへ速度変更
ACS2002/TIM/1500/SPD/6000:前変更点から1500msのところで10000PPSへ速度変更
ACS2003/TIM/2000/SLW:前変更点から2000msのところで減速停止へ
ACS2004/END:ﾌﾟﾛｸﾞﾗﾑの終了(必ず書き込む）
ACS3000/TIM/0/SPD/1000 (ﾀﾞﾐｰ) :ｽﾀｰﾄ点から0msのところで1000PPSへ速度変更
ACS3001/TIM/1000/SPD/3000:前変更点から1000msのところで6000PPSへ速度変更
ACS3002/TIM/1500/SPD/6000:前変更点から1500msのところで10000PPSへ速度変更
ACS3003/TIM/2000/SLW:前変更点から2000msのところで減速停止へ
ACS3004/END:ﾌﾟﾛｸﾞﾗﾑの終了(必ず書き込む）
最初の変更タイミングを時間にして、複数のモータを同時スタートする場合はダミーの変更タイミングを先頭番地に用意しておく必要があります。
（スタート準備ができた段階からタイマがスタートするので。
2番目のタイマは前の変更が開始されてからのスタートになります）
２．データリードコマンド
REMOTE/LOCAL双方で使用できます｡
データ確認用コマンドです。
書き込まれているデータを確認できます。
ACS?xyyy
reply:xyyy/PNT/JJJ･･･J/FNC/LLL･･･L
前項の設定をした場合の読出し例）
送信コマンド
ACS?0000
ACS?0001
ACS?2000
ACS?2001

応答内容
0000/ADD/005000/SPD/003000
0001/ADD/010000/SPD/006000
2000/TIM/000000/SPD/001000
2001/TIM/001000/SPD/003000

6桁以下のデータは6桁で返されます。
6桁以上のデータは必要桁分返ります。

３．オートチェンジスキャン準備コマンド
リモートモードで使用できます。
複数軸の同期スタートを行うときは軸毎に与えます。
オートチェンジスキャン開始前に必ず必要です。
ACSPx オートスキャンデータをセットします。
データがセットされたことを次項のコマンドで確認できます。
最後の変更点の次にデータの終了を示すデータ(END)が書かれていないとセットできないのでスキャン準備完了フラグも立ちません。

- 19 -

４．オートチェンジスキャン準備確認コマンド
REMOTE/LOCALモードで使用できます。
スキャン準備を確認します。
ACSP?x"reply:ready, not ready"動作開始すると、not readyになります。
データ書き込みコマンドを実行するとそのチャンネルはnot readyになります。
not readyの状態ではAUTO CHANGE ドライブにはなりません。
５．オートチェンジスキャン準備完了クリア
リモートモードで使用できます。
スキャン準備をクリアします。
ACSCx
コマンドによりx チャンネルは not readyになります。
既に準備したAUTO CHANGE ドライブを中止したいときに使用します。
not readyの状態ではAUTO CHANGE ドライブにはなりません
６．同期スタート用PAUSE ON
リモートモードで使用できます。
複数軸の同期スタートを行うときのみ使用します。
PAUSE ON or S3016
７．(オートチェンジ)スキャンコマンド
リモートモードで使用できます。
複数軸の同期スタートを行うときは軸毎に与えます。
以下の4つのコマンドはオートチェンジスキャン準備が完了しているとき
オートチェンジスキャンコマンドとして機能します。
ABSx(B)±dddddddd
RELx(B)±dddddddd
SCANPxSCANNx絶対位置移動相対位置移動＋方向移動－方向移動

８．同期スタート用PAUSE OFF
リモートモードで使用できます。
複数軸の同期スタートを行うときのみ5．項のPAUSE ONとペアで使用します。
PAUSE OFF or S3017
オートチェンジスキャンコマンドで移動中のモータは、通常の停止コマンドで途中停止できます。

- 20 -

９．オートスキャンコマンド使用時のタイミング誤差について
オートスキャンコマンドを使用したとき、変更開始条件を検出してから変更が開始されるまでのタイミングは一定速中と、加減速中で異なり以下のようになります。
一定速中： < 320μs
加減速中： < 160μs + 検出時の変速周期
A pos と B pos、C pos と D pos に同時に変更開始条件が検出されたとすると、
A,C posのタイミングより B,D posのタイミングは80μs遅れます。
4軸同時駆動の場合は注意が必要です。
変速周期データ（抜粋） 単位：ms
RATE No
ms/KHz
L1 type01,000.000
50.0007510.000
25.50017200.000
10.00024
100.000
5.000
31
51.0002.550
4120.000
1.000
55
5.100
0.255
60
3.300
0.165
67
1.600
74
0.820
76
0.680
82
0.390
84
0.330
89
0.200
91
0.160
98
0.082
100
0.068
106
0.039
113
0.020
115
0.016

L2 teyp

50.00025.00012.750
5.000
1.275
0.825
0.400
0.205
0.170

M1 type50.00025.50010.000
2.550
1.650
0.800
0.410
0.340
0.195
0.165

M2 type51.00020.000
5.100
3.300
1.600
0.820
0.680
0.390
0.330
0.200
0.160

H1 type50.00012.750
8.250
4.000
2.050
1.700
0.975
0.825
0.500
0.400
0.205
0.170

H2 type51.00033.00016.000
8.200
6.800
3.900
3.300
2.000
1.600
0.820
0.680
0.390
0.200
0.160

１０）その他のコマンド
VER?
内蔵ファームウェアのバージョン情報読出しコマンドです
応答は 1.00 06-10-14 PM16C-04X などとなります。
(PM16C-04XD,04XDLともファームウェアは共通バージョンを使っています)ACSK?x

オートチェンジスキャンデータがパルス出力コントロールIC MCC06に書かれた状態を読出せます。変更点を示すデータです。
reply
aabb,hhhh,hhhh
aa:code(00:END,01:ADD,02:TIM,80:ACC,81:DEC)
bb:sequence number(0 to 7F)
hh･･･hh:data

ACSH?x

オートチェンジスキャンデータがパルス出力コントロールIC MCC06に書かれた状態を読出せます。変更内容を示すデータです。
reply
aabb,hhhh,hhhh
aa:code(00:SPD,01:RTE,02:SLW,03:FST,0F:NOP)
bb:sequence number(0 to 7F)
hh･･･hh:data

ERR?

エラー状態読み出しコマンド
複数のエラーがあるときは下位ビットのエラー情報を返信します。

- 21 -ERRF?

エラーフラグ読み出しコマンド（8ビットを16進数で返信します）
B0:COMMAND ERROR, B1:MCC06 BUSY ERROR, B2:BAD ABS COMMAND

ERRC

全エラー状態クリア

ERRCx

指定エラーのみクリアします。
x:0 COMMAND ERROR クリア:1 MCC06 BUSY ERROR クリア:2 BAD ABS COMMAND クリア

FROM?

保守用コマンドです。二つのフラッシュROMのうちどちらが使われているかを読出します。FROM0,FROM1などと読み出せます。

FROM1,FROM2

保守用コマンドです。二つのフラッシュROMのうち、使用する側を指定します。
電源の再立ち上げや"REST"コマンドで指定した側のファームウェアで全てのパラメータを初期化して起動しますので、注意が必要です。

REST

保守用コマンドです。電源のOFF/ON手順を使わないでリセットスタートするコマンドです。

STEM?

保守用コマンドです。MCC06のstatus port1の内容を読み出します。
aaaa/bbbb/cccc/dddd のようにA,B,C,D control IC のstatusが16進数で出力
されます。

VERH?

保守用コマンドです。ハードウェアバージョンを読み出せます。
HD-VER0,HD-VER1,HD-VER2,HD-VER5, HD-VER6などと読み出せます。
HD-VER0:PM16C-04XD, HD-VER1:PM16C-04XD, HD-VER2:PM16C-04XDL
HD-VER5,6:PM16C-04XD,-04XDL/DIR変更可ハード搭載版を表します。

１１）追加コマンド・機能(V1.14～)
SETCHabcd
任意のチャンネル(0～9,A,B,C,D,E,F)をコントロールウィンドウの
A,B,C,Dpositionにセットします。
SETCH01-- などと"-"を使うとそのチャンネルのみ変更されません
リモートモードでのみ有効です。変更対象チャンネルが動作中などでbusyの時はコマンドは無視されます。
SETCH?

現在のチャンネル読み出しコマンドです
応答は "34AB" などとA,B,C,Dposition順に返されます。

(V1.14～)
ハンドボックスの速度変更ボタンでモータを停止できる機能
リモート制御中にハンドボックスの速度変更ボタンで動作中のモータがスロー停止できます。
このボタンの長押し3秒で停止可能モードに入り、H,M,L LEDの全点灯になります。
モードから抜けるのも3秒です。
(V1.39～)駆動16チャンネルそれぞれに半角英数字3文字のコメントを設定できます。
CHCMNT?x
各チャンネルを指定してコメントをリードできます。
x=a で全チャンネルのコメント文字をリードできます。
CHCMNTx ***指定したxチャンネル に3文字のコメント*** を設定できます。
LCD_DISP T
LCD表示 TimingOut ポートを表示します。
LCD_DISP C
LCD表示 コメントデータを表示します。

- 22 -

LCD画面のバックライトの明るさをコントロールします。
LCDB?
データリード 返信は LCD BRIGHT 100
LCDB 100
1～100の値で明るさを調整します。

1～100の値を返します。

(V1.47～)"STS?"コマンドの1チャンネル版です。
STSx?
指定したチャンネルxの詳細データを読み出します。
応答はR(L)aPVHH±uu･･･uの形式で、詳細は"STS?"コマンド(5ページ)を参照して下さい。
(V1.48～)ホールドオフ信号出力時、モータ動作前にホールドオフ信号を切ってから動作するまでの時間を変更します。
HOLDTM?x
チャンネルxの動作前ホールドオン時間を読み出します。
応答は dddms (ddd: 50～500 単位:msec.)です。
HOLDTMxddd
チャンネルxの動作前ホールドオン時間を設定します。
dddには50～500までの値を10刻みで入力します。 (単位: msec.)- 23 -

(V1.43～)
揺動動作モードを追加しました。
B Position(以下、B軸)に割り当てられたモータを往復動作させ、1往復する度にA軸をステップ移動させて再びB軸を1往復させる動作をA軸が一定回数動くまで繰り返すモードです。
A,B軸を使う2軸動作モード、A,B,C軸を使う3軸動作モードが主で、設定により、A軸と同じタイミングで起動するD軸も使用可能です。
この揺動動作モードでは計測開始時より内部タイマによる計測を行っており、いずれのモードにおいてもB軸の周期を一定にした動作を行い、A軸の動作周期の読出しが可能です。また、タイミングパルス出力機能を併用する事でB軸の TTL OUTPUT(TP1) よりB軸の動作に対応してGATEまたはPULSE信号を出力できます。
･2軸動作モード
下図は、A軸動作回数を4にした時の動作です。最初にB軸が1回往復動作を行い、停止後
5～10msec程でA軸（D軸同期モード時はD軸、B軸ジャンプモード時はB軸も）が起動します。
A軸停止後5～10msec程で再びB軸が起動し、設定された回数の動作を行ったら終了します。

2軸動作モード（B軸 ジャ ン プ モ ード 有 効 時）

2軸動作モードD軸(D 軸 同 期 時)
A軸

D軸
(D 軸 同 期 時)
A軸

B軸

B軸

時間
GATE
PULSE

時間
GATE
P ULSE

1周期

1周期･3軸動作モード
下図は、A軸動作回数を2、C軸動作回数を3にした時の動作です。2軸動作モードとの違いは、
B軸が1往復した後もB軸が設定回数動作していない時はA軸が動作せず、代わりにC軸が動作します。
設定回数動作した時にC軸は最初の位置に戻り、A軸が動作します。
（D軸同期モード時はD軸、B軸 ジャンプモード時はB軸も動作）

3軸動作モード
A軸
D軸（D軸同期時）
C軸

B軸

時間
GATE
PULSE
1周期

- 24 -

3軸動作モード（B軸ジャンプモード有効時）

A軸
D軸（D軸同期時）
C軸

B軸

時間
GATE
PULSE
1周期

･信号出力
タイミングアウトモードを設定している場合、前述の図のGATEまたはPULSEのタイミングで信号が出力されます。以下のタイミング信号出力設定コマンドで設定をして下さい。
TMGMxY

x:B軸に割り当てられたチャンネル (0～F)
Y:信号設定 0/disable(出力なし) 1/GATE
2/200ns 3/10μsec 4/100μsec 5/1msec PULSE

TMGM?xコマンドで設定を読出せます。

(返答は0～5)･揺動動作関連コマンド一覧（R)はリモートモード、(R/L)はリモートローカル両方で有効です。
揺動動作時のパラメータを設定するコマンドは、揺動動作中（中断を含む）は無視されます。
必ず揺動動作を停止している状態で設定を行って下さい。
YMDabc(R)

揺動動作モードでの動作を設定します。
a: 2軸揺動動作モードか3軸揺動動作モードを設定します。
2/2軸動作モード 3/3軸動作モード
b: A,D軸同期動作モード D軸がA軸と同じタイミングで起動します。
0/無効 1/有効
c: B軸ジャンプモード A軸が起動するのと同時にB軸もステップ移動します。
0/無効 J/有効

YMD?
(R/L)

現在の揺動動作モード及び動作状態を読出します。
返答は「YMDabc xxxx」で、 xxxxには現在の揺動動作の状態に応じてSTART/STOP/PAUSEのいずれかが入ります。

YSTRT
(R)

揺動動作を開始します。揺動動作で使用する軸が動作中、移動後のパルス値がオーバーフローするような場合ですと動作しません。
揺動動作中は元の設定にかかわらずホールドオフ機能が無効になります。
揺動動作を停止しますと元に戻ります。

YPAUS
(R)

揺動動作を中断します。もう一度このコマンドかYSTRTを送信しますと
揺動動作を再開いたします。
周期計測の関係上、A軸の動作回数が2になる前に中断しますと
正確な計測が不可能になり、正常に動作しなくなる恐れがございます。

- 25 -

YSTOP(R/L)

揺動動作を停止します。揺動動作で使用している全ての軸をパネルのSTOPボタンを押したときと同じ停止動作で停止させます。

YSTS?
(R/L)

現在のA軸、C軸の動作回数を読出します。
返答は「RPaaa CRccc」で、aaaにはA軸の動作回数、cccにはC軸の動作回数が3桁以上の整数で入ります。
2軸動作モード時cccは常に000です。

YCY?
(R/L)

A軸の動作周期を読み出します。返答は4桁以上の整数で、単位はmsecです。
1回もA軸が動作していない場合は0000が返答されます。

YAP±d..ddd(R)

A軸のステップ移動パルス量を設定します。
±d...ddd: -2,147,483,647～+2,147,483,647

YAP?
(R/L)

A軸のステップ移動パルス量を読出します。
返答は符号付7桁以上の整数です。

YBP±d..ddd(R)

B軸の移動パルス量を設定します。

YBP?
(R/L)

B軸の移動パルス量を読出します。
返答は符号付7桁以上の整数です。

YRPd..ddd(R)

揺動動作の動作数(A軸の動作回数)を設定します。
d...ddd: 1～2,147,483,647

YRP?
(R/L)

揺動動作の動作数(A軸の動作回数)を読出します。
返答は3桁以上の整数です。

YBJ±d..ddd(R)

B軸ジャンプモード時のステップ移動パルス量を設定します。

YBJ?
(R/L)

B軸ジャンプモード時のステップ移動パルス量を読出します。
返答は符号付7桁以上の整数です。

YCP±d..ddd(R)

C軸のステップ移動パルス量を設定します。

YCP?
(R/L)

C軸のステップ移動パルス量を読出します。
返答は符号付7桁以上の整数です。

YCRd..ddd(R)

C軸の動作回数を設定します。
d...ddd: 1～2,147,483,647

YCR?
(R/L)

C軸の動作回数を読出します。
返答は3桁以上の整数です。

YDP±d..ddd(R)

D軸のステップ移動パルス量を設定します。

YDP?
(R/L)

D軸のステップ移動パルス量を読出します。
返答は符号付7桁以上の整数です。

- 26 -(桁数は任意以下同)･コマンド使用例
あらかじめ、それぞれの軸でSPEED、RATE、揺動動作させたいモータ、初期位置の設定を行っているものとします。

1/B軸の移動パルス量を1000、A軸のステップパルスを50、動作回数を100とした
2軸揺動動作を行う場合
YMD200
YAP50
YBP1000
YRP100YSTRT

2軸揺動動作モードに設定します。
A軸のステップパルスを50に設定します。
B軸の移動パルスを1000に設定します。
動作回数を100に設定します。
揺動動作を開始します。

2/B軸の移動パルス量を2500、A軸のステップパルスを120、C軸のステップパルスを100、C軸の動作回数を5、動作回数を50とした3軸揺動動作を行う場合
YMD300
YAP120
YBP2500
YCP100
YRP50
YCR5
YSTRT

3軸揺動動作モードに設定します。
A軸のステップパルスを120に設定します。
B軸の移動パルスを2500に設定します。
C軸のステップパルスを100に設定します。
動作回数を50に設定します。
C軸の動作回数を5に設定します。
揺動動作を開始します。

3/B軸の移動パルス量を-1000、B軸のジャンプパルスを10、A軸のステップパルスを10、D軸のステップパルスを20、動作回数を200とした2軸D軸同期揺動動作を行う場合(移動、ステップパルスには0や負の数も設定出来ます。B軸の移動パルスを負に設定した場合、CW → CCW 移動が CCW → CW 移動になります。0の場合は移動しません。)
YMD21J
YAP10
YBP-1000
YBJ10
YDP-20YRP200
YSTRT

2軸、D軸同期、B軸ジャンプ揺動動作モードに設定します。
A軸のステップパルスを10に設定します。
B軸の移動パルスを-1000に設定します。
B軸のジャンプパルスを10に設定します。
D軸のステップパルスを-20に設定します。
動作回数を200に設定します。
揺動動作を開始します。

･GATE、PULSE信号について
B軸の動作に合わせて出力される
GATE信号、PULSE信号ですが、B軸が戻ってきた時の信号のタイミングは厳密には若干差異があります。
GATE信号はB軸が停止点に到達した時に立ち下がりますが、PULSE信号はB軸が停止点の1つ手前に到達した時に信号を出力します。

拡大図

停止点+1
A

B

GATE
PULSE

- 27 -

４．ファームウェアのバージョンアップ
PM16C-04XD(L)はファームウェアのバージョンアップが通信環境を使って行えます。
ファームウェアのバージョンアップ時には安全のためPM16C-04XD(L)とドライバとの接続を切るか、ドライバの電源を切った状態で行ってください。大まかな手順は以下の通りです。
RS232CやLANをお使いいただくと便利です。
ツジ電子のホームページから専用プログラム(Windows用)をダウンロードしてお使いください。
http://www.tsuji-denshi.co.jp/download_file/lan_rs_file_send.EXE(専用プログラム本体)
http://www.tsuji-denshi.co.jp/manual_pdf/pm16c_04xd_vup_soft.pdf(PDF形式 取扱説明書)
ここではフリーソフトのTeraTermでLANを使ってインストゥールする方法で説明します。
① バージョンアップ用のテキストファイルをツジ電子のHPからダウンロードし、解凍します。
② TeraTerm を起動します。
TCP/IPを選択しPM16C-04XD(L)のIPアドレスとポート番号を入力します。
SETUP -> TerminalでNew-lineの設定をReceive Transmit 共に CR+LF にします。
Local echoにチェックマークを入れて OK をクリックします。
VER?と入力して応答があれば正しく接続されています。
③

PM16C-04XD(L)をREMOTE MODE にします。
パネルから手動で行っても TeraTermから"REM"コマンドで行ってもかまいません。

④

TeraTermの File -> Send fileをクリックするとファイル選択窓が開きますので、①で用意したファイル(拡張子が.Sのファイル)を指定し、開くボタンをクリックするとダウンロードが開始されます。

⑤

TeraTermによりファイルがPM16C-04XDに送られる様子がパソコンの画面でご覧になれます。
PM16C-04XD(L)側では受信中のサイン"REMOTE"ランプがゆっくり点滅します。

⑥

約90秒でダウンロードが終わると"REMOTE"ランプが速い点滅に変わり、ROM書き込みを開始します。
約20秒で書き込みが完了すると"REMOTE"ランプは消灯し"LOCAL"ランプが点灯して、書き込み完了を知らせます。

⑦

TeraTermを終了してTCP/IPの接続を切ってからPM16C-04XD(L)の電源を切り、再びPM16C-04XD(L)の電源を入れるとバージョンが新しくなって立ち上がります。
電源を切りたくないときは、"REST"コマンドで同じことができます。(V1.09～)

もし、ダウンロードの途中でアクシデントによりダウンロードが完全に行えなかった場合にはPM16C-04XD(L)電源の再投入と同時に暴走する（初期画面が表示されない）ことがあります。
この場合には、一度電源を切ってテンキーの"0"(XDL型は"8")を押しながら電源を入れると、元のバージョンで立ち上がりますので再度、インストゥールを行ってみてください。
（この方法は通常でも以前のバージョンに戻したいときなどにも使えますが、
ローカル画面のファームウェアを表す"NEW", "OLD"表示は変わりません）
元のバージョンでも立ち上がらなくなったら内蔵ROMに古いバージョンが入っているのでこれを立ち上げて回復できます。以下にその手順を示します。
内部操作をする場合は必ず、コンセントを抜いて感電しないように注意して行ってください。
不安がある場合は、弊社にお申し付け下さい。

1．電源を切って上蓋を開け、プリント基板 "TEP036"(XDL型は"TEP063")のボタン電池そばにある
ディップスイッチの2をON側にします。(1はOFFのままです）
2．前面パネルのテンキーの "5" を押しながら電源を入れると、REMOTEを示す赤LEDが約15秒点滅し、その後赤LEDの点灯状態になって終了します。
この時点では表示窓は正しい動作を行っていません。

- 28 -

3．再び電源を切ってディップスイッチの2をOFFにします。（1はOFFのままです）
4．REM/LOCボタンを押しながら電源を再投入するとV1.00のファームウェアで立ち上がります。
（LCD表示が現れないこともありますが、REM/LOCボタンによる切換ができればOKです)
ここから先はもう一度最新のファイルを上記①から⑦の手順で書き込んでください。
V1.00のファームウェアではIPアドレスが[192.168.1.55]、ポート番号が[7777]に、RS-232C通信時のボーレートが[9600]に初期化、固定されます。
パネル面からセットアップモードに入りP8のPROGRAM CHANGEにより、ファームウェアのNEW <-> OLDの切換ができます。古いバージョンでの動作を試してみたい
場合などにこの機能をお使い下さい。
なお、ファームウェアの入れ換え時には位置データは全てゼロになり、設定値類は全てデフォルト状態になります。
５．改訂履歴2007.09.13(改訂8)"PAUSE ON", "PAUSE OFF" コマンド追加(V1.13～)
走行中に予定に従った速度変更などを可能にするオートチェンジドライブ機能を追加(V1.13～)
2008.07.07(改訂9)STOPMDxABのAB順番違いを取説上で修正(→V1.14～)"SETCHabcd", "SETCH?", "HDSTLS?"コマンドを追加(→V1.14～)
リモートモードでも自動ホールドオフ機能が働くようにした(→V1.15～)
2009.09.16(改訂10)
HOLD ON/OFF コマンド "HOLDxON", "HOLDxOFF", "HOLD?x"/reply:"ON" or "OFF" 追加(→V1.17～)
V1.20～V1.29：skipped
PULSE-DIR 出力設定時、DIRの論理を変更できる("VERH?"/"HD-VERn" n > 4)(→V1.30～)
リモート制御時にハンドボックスの速度変更ボタンで、動作中のモータを停止できる(→V1.31～)2011.03.01(改訂11)
Teratermのバージョンによって通信がうまくいかないと言う指摘を受けて、ファームウェアのバージョンアップ用のプログラムをHPにアップロードし、
4.ファームウェアバージョンアップの項にに加えた。LCDバックライト明るさの制御と、各チャンネルにコメント文字を（3桁）入力できるようにした。
2011.06.29(改訂12)
モータが停止した場合、GPIBラインのSRQラインをアクティブにしていたが、加えてモータ停止時にRS232Cライン、LAN通信からも "STOPx"を出力するようにした。
2011.11.10(改訂13)揺動動作モード(P24～27)の追加、揺動動作関連(Y～)のコマンドを追加
2013.05.17(改訂14)
V1.47に追加された指定の1チャンネルの詳細を読出すSTSx?コマンドの追加
2013.06.19(改訂15)
V1.48でモータ動作時にホールドオフを解除してから動作をするまでの時間を変更できるコマンドの追加
本文の体裁を調整- 29 -

2014.06.03(改訂16)
パネルからIPアドレスおよびポート番号を設定可能な為、LAN通信設定の項目で従来の設定方法を削除
2017.10.13(改訂17)
内蔵ROMからV1.00のファームウェアを立ち上げた時、IPアドレス、ボーレートが初期設定に戻り、固定されることを追記- 30 -

６．性能・仕様
電源

入出力

LCD表示器

ﾊﾟﾈﾙ押ﾎﾞﾀﾝと表示ﾗﾝﾌﾟAC 85V ～ 264V

47 - 440Hz

制御ﾓｰﾀ数

最大16台

同時制御ﾓｰﾀ数

最大 4台

制御出力

各ﾓｰﾀﾄﾞﾗｲﾊﾞに対してCW, CCW, HOLD OFF信号 5V 8mA(ﾗｲﾝﾄﾞﾗｲﾊﾞ)

出力周波数1～5MPPS

制御ﾊﾟﾙｽ数

0～±2,147,483,647

加減速ﾚｰﾄ1048.56～0.0125 ms/kHz

加減速形式

S字、台形

出力仕様

2ﾊﾟﾙｽ/1ﾊﾟﾙｽ

ﾊﾟﾙｽ出力ｺﾈｸﾀ

DE9S(F)/PM16C-04XD, RJ-4/PM16C-04XR

ﾘﾐｯﾄSW入力

各制御ﾓｰﾀに対して
CW-LS, CCW-LS, HOME-LS
ｾﾝｻｰ用電源+12Vも用意されています(MAX 1A)

ﾘﾐｯﾄ入力ｺﾈｸﾀ

DE9P(M)/PM16C-04XD(L), RJ-4/PM16C-04XR(L)LCD表示器仕様

PM16C-04XD : 40文字×4行(文字ｻｲｽﾞ 4.9H*2.8W)
PM16C-04XDL: 20文字×4行(文字ｻｲｽﾞ 9.2H*4.8W)

通常表示

ﾁｬﾝﾈﾙ表示、±10桁の現在位置、速度表示、ﾘﾐｯﾄSW情報
相対位置移動値、絶対位置移動値、ﾌﾟﾘｾｯﾄ値、原点位置情報
パネル操作ready/not ready、

設定表示

ﾘﾐｯﾄSW設定情報、ﾊﾟﾙｽ出力ﾓｰﾄﾞ、HSPD,MSPD,LSPD値、加減速ﾚｰﾄ
LS､PB停止ﾓｰﾄﾞ、原点検出機能、ﾀｲﾐﾝｸﾞﾊﾟﾙｽ発生機能、RS232C設定、GP-IB設定、HANDBOX選択、FIRMWAREﾊﾞｰｼﾞｮﾝ選択

PM16C-04XD

ｽｲｯﾁ･ﾎﾞﾀﾝ：JOG CW/CCW, PRESET,START,STOP,REM/LOC,SETUP
REM/LOC,CURSOR,CHG,MODE,TEN-KEY
ﾗﾝﾌﾟ：REM/LOC,START,STOP,SETUP,HP/REL/ABS/SCANﾗﾝﾌﾟ

PM16C-04XDL

PM16C-04XDに加えてｽｲｯﾁ･ﾎﾞﾀﾝ：CH-RDY*4,CH-SEL*4,DSP-CHG
ﾗﾝﾌﾟ：CH-RDY*4,CW_CCW_PULSE*4,CW_CCW_HP_LS*4,HLDOFF*4

ﾃﾝｷｰとｶｰｿﾙ

ｺﾝﾄﾛｰﾙﾎﾞﾀﾝ

歩進ﾓｰﾄﾞ

50VA

12V 5mA(-ｺﾓﾝ)

ﾃﾝｷｰとｶｰｿﾙでﾁｬﾝﾈﾙ選択、4つのﾁｬﾝﾈﾙのｺﾝﾄﾛｰﾙ ready/not readyの選択、設定値の変更ができます。ﾁｬﾝﾈﾙ毎にH,M,Lの速度選択ができます。H,M,Lの速度はﾁｬﾝﾈﾙ毎に任意に設定できます。
PRESET
START
STOP
JOG

ready chの位置表示器に数値設定器の内容をﾌﾟﾘｾｯﾄ
ready chのﾎﾟｼﾞｼｮﾝの制御を歩進ﾓｰﾄﾞに従ってｽﾀｰﾄ
ready chのﾎﾟｼﾞｼｮﾝの制御をｽﾄｯﾌﾟ
ready chのﾎﾟｼﾞｼｮﾝJOG歩進SCAN MODE

ready chを連続歩進します。方向はJOG SWによります。

ABS IDX MODE

ready chを設定値まで移動します。

REL IDX MODE

ready chを設定値だけ移動します。

HP STOP MODE

HOME POSITION LS で停止します。
動作形態は原点検出記憶が無い場合2種類、原点検出記憶がある
場合1種類の計3種類あります。

通信機能LAN, GP-IB, RS232C通信にてﾘﾓｰﾄｺﾝﾄﾛｰﾙできます。

ｹｰｽ

EIA 2 UNIT ﾗｯｸﾏｳﾝﾄ型(88H×482W×325D)
その他ご不明の点は、下記宛お問い合わせください
ツジ電子株式会社 開発・設計部〒300-0013 茨城県土浦市神立町3739
TEL 029-832-3031(代) FAX 029-832-2662
E-mail : info2@tsuji-denshi.co.jp- 31 -