Quad Load Cell Interface 2 - QLCI2
Data sheet

LINAK.COM/MEDLINE-CARELINEQLCI2The Quad Load Cell Interface 2 (QLCI2) for the hospital andcare segment is a scale system accessory with weighing
capabilities and Out of Bed functionality.
The housing makes the QLCI2 easy to mount by uniqueslide-on brackets and has an IPX6 Washable DURATM ingress
protection.
The QLCI2 supports the LINAK® OpenBusTM system, offeringa high level of customisation.
LINAK A/S delivers the OpenBusTM system only and is not
responsible for any products (i.e. products from 3rd partysuppliers) other than LINAK products or the compatibility ofsuch products with the LINAK OpenBusTM system.

Features and options:• Housing colour:
Light grey, RAL 7035
• Protection class:
IPx6 Washable DURATM• Weight:
406 g• Power consumption:
V-Permanent 40 V = < 50 mA		
V-Bus 8 V = < 15 mA• Mounting:
Mounting bracket or bed frame• Cable lock:Integrated in lid• OpenBus interface:
1 port - 10-pin modular plug
• Calibration interface:
1 port - 6-pin Micro-Fit• Load cell interface:
4 ports - 6-pin Micro-Fit• Optional relay gateway: 6-pin Minifit with two		relays NC-NO• Standby power:
There is no standby consumption		
if the system is on sleep• Power supply:
Via OpenBusTM• Operation mode:
On mains or on battery

LINAK.COM/MEDLINE-CARELINE

Usage:• Operation temperature:• Storage temperature:• Relative humidity:• Atmospheric pressure:• Height above sea level:• Approvals:										• Compatibility:• Flammability rating:• Latex free:+5 °C to + 40 °C-10 °C to + 50 °C
20% to 80% - non-condensing
700 to 1060 hPa
Max. 3000 meters
IEC60601-1
ANSI/AAMI ES60601-1
CAN/CSA-22.2 No 60601-1
EN 45501 / OIML R76
EU type examination according
to 2014/31/U
All OpenBus control boxes
UL V0YesQLCI2Ordering example:TYPE

QLCI2

Not used00

00 = not used

Not used

00 = not used

SW option

00 = Standard software
X = Special software

Relay option

0

0 = Standard (2 relays)

Prepared for relay accessA

A = Yes

QLCI2

B = NoHW functionality

1

1 = Standard version
X = Special functionality

Housing / Colour

22 = RAL7035 (light grey)

IP protection class

99 = IPX6 Washable DURATMCODENUMBER (FOR PRICE CALCULATION):QLCI200000A129

MA-S8-00-255-A

Prices can be subject to change without prior notice1

QLCI2 - System Overview

HB100

CO61

MJB5BL1LA40

QLCI24x Load Cell
3rd party

BL1LA40

Gateway
The Gateway port provides an interface from the OpenBus™ to a 3rd party hardware, for instance a nurse call system. The Gateway hardware isnot a LINAK product and must be supplied by a 3rd party supplier.
3rd party products specifications must be in accordance with the QLCI2 interface description and isolation requirements.

CablesGateway cableItem no.: 1013W7008-X
Colour: RAL 7035
Material: PVC

Drawing No.: 1013W7008

Calibration cable
Item no.: QLCI2CALCABLE-00

Heat shrink tube

Drawing No.: 1013W7003

Dimension drawing:

Drawing no.: 1013w4020

Mounted on frame:
Service lid
Integrated hinge
Removable if needed

Release hook

3-way cable exit

Easy-mount bracket

Very small built-in dimension:
Height: Min. 90 mm (lid can be removed)
Length: +20 mm (locking system)

Drawing No.: 1013W4008

Mounting bracket (frame flat)- Article No. 1015W1001:

Mounting bracket (frame flat) w/M4 nuts
- Article No. 1015W9009:

Drawing No.: 1015W4001

Drawing No.: 1015W1009

It is recommended to mount the QLCI2 in a position that allows water to escape.
Recommended torque: 0.6 Nm +/-0.1
The bracket can be mounted to the bed frame or any other application by means of one of the following mounting procedures:
1) M6 nut to be placed in bracket and fixed with M6 bolt from the rear side.
2) M5 machine screw with flat washer to be fixed through bracket with nut on the rear side.
3) Self-tapping screw to be placed through bracket and onto the frame.

Mounting of cables and cable lock:
The QLCI2 have a uniquely designed cable lid. The lid also works as an integrated cable lock when closed.
1. Mount load cell cable plugs in QLCI2
2. Calibrate the system
3. Mount blind plug in calibration port
4. Close lid until lock snaps into place
5. Place calibration void label - only one is needed
To allow free access to the cables, the lid has a rest position when completely opened.
It is possible to remove the lid by lifting it a few degrees and pull it away from the housing under tight mounting conditions.

1

Mount load cell cables

2

Calibrate the system

4

Close lid until it snaps

5Place void label

Only 1 label is required. The label can be placed
anywhere along the opening of the lid on theQLCI2 box.

3

Mount blind plugDisplay options
HB100 Weighing Solution
The HB100 is an intelligent hand control with the LINAK® Weighing Solution,
capable of displaying the weight of a patient as well as setting the Out Of
Bed feature.
HB100 comes with a 2.4” colour display and a replaceable cable.

Scale verification
It is possible to do a verification of the scale system in 1g. resolution and with the option of enabling the destination gravity. The scale verification
section is placed under settings for the HB100.
It is also possible in Service Data Tool. The scale verification menu is placed under QLCI view.

Calibration
To calibrate the QLCI2 a computer with the calibration software installed is needed. To connect the computer and the QLCI2, a QLCI2 Calibration
cable is needed.

OpenBusTM cable

COXX

Calibration cable

QLCI2

LINAK
Calibration
softwareCopyright © LINAK 2020.06. MA M9-02-768-G . Chapter 10.50
Terms of use
The user is responsible for determining the suitability of LINAK products for specific application. LINAK takes great care in providing
accurate and up-to-date information on its products.
However, due to continuous development in order to improve its products, LINAK products are subject to frequent modifications andchanges without prior notice. Therefore, LINAK cannot guarantee the correct and actual status of said information on its products.
While LINAK uses its best efforts to fulfil orders, LINAK cannot, for the same reasons as mentioned above, guarantee the availabilityof any particular product. Therefore, LINAK reserves the right to discontinue the sale of any product displayed on its website or listed
in its catalogues or other written material drawn up by LINAK.
All sales are subject to the Standard Terms of Sale and Delivery for LINAK. For a copy hereof, please contact LINAK.

FOR MOUNTING INSTRUCTIONS AND GUIDANCE IN USAGE, PLEASE SEE THE RELEVANT USER MANUAL