Dry Helium Leak Detector UL5000

ULTRATEST™ Sensor Technology

Defining Speed and Accuracy in Leak Detection
The INFICON UL5000 Helium Leak Detector is designed to meet themost critical and demanding leak detection applications. Featuring
INFICON proprietary software algorithms I-CAL and Hydro-S in a fieldproven vacuum design, the UL5000 provides testing flexibility, highsensitivity and quick accurate results making any leak detection
application fast and easy.
The UL5000 delivers fast response times in all measurement ranges
and extremely short cycle times in reaching test conditions and finalresults. The specially designed vacuum architecture provides thecontinuous high helium pumping speeds and the extremely fastresponse times you demand.

1Dry Helium Leak Detector UL5000

FEATURESHYDRO-S (HYDROgen-Suppression) to enable test
conditions to be reached quickly.

I-CAL (Intelligent - Calculation Algorithm for Leak Rates) toensure fast response to leaks in all measurement ranges.

A special multiple inlet turbomolecular pump that provides
optimized high helium pumping speeds with high helium
compression at all pressures to deliver excellent sensitivity,response, and fast clean-up.

A booster turbomolecular pump for extremely fast responseon any test object, including the largest volumes andchambers.

The field-proven INFICON magnetic sector mass
spectrometer for years of accurate and repeatable results ina trouble-free, maintenance-free design.

Self-protection features to protect the UL5000 from helium
and particulate contamination and even an auto-purge cycle
to ensure the UL5000 is ready when you need it.

A rotatable display / control interface and optional remote
hand controller for operational flexibility.

Access control to the unit and system software can be
enabled via the software to prevent unauthorized use orunintended modifications to test set-ups.

Software updates via e-mail to keep your UL5000 at
optimum performance.

Work Station Design with an optimal height work surface
that includes an ESD mat and tool box for testing
convenience.

ORDERING INFORMATIONTypeUL5000 Helium Leak Detector

Leak Detector UL5000, 100V / 115 V

550-501A

UL5000, 230V AC, EU plug

550-500A

2Dry Helium Leak Detector UL5000

SPECIFICATIONSTypeUL5000 Helium Leak DetectorMin. detectable leak rate for
Helium ( Vacuum mode )mbar•L/s<5E-12

Min. detectable leak rate for
Helium (Sniffer mode )mbar•L/s

8E-8

Max. Inlet pressure GROSS
mode

mbar

15

Max. Inlet pressure FINE mode

mbar

2

Max. Inlet pressure ULTRA
mode

mbar

0.4

Pumping speed during
evacuation

m³/h

25 at 50 Hz

Helium pumping speed GROSS L/s
mode

max 8

Helium pumping speed FINE
mode

L/s

max 20Helium pumping speed ULTRA
mode

L/s

>20

Detectable masses

2,3,4 amu

Mass spectrometer Sector Field

180 °C

Filaments ion sourceIridium/Yttria coated

2Calibrated built-in test leakE-7

Test port

DN40KF

Adjustable triggers

2Interface

RS232

Chart recorder output

2 x10 V

In/Outputs

PLC compatible

Permissable ambient°C
temperature ( during operation )Type of protection+10...+40
IP 20

Weightkg

140

Weightlb.

308

Dimensions ( LxWxH )mm

1068x530x1083

Dimensions ( LxWxH )in.

42.5x21x42.6

Supply voltage

V (ac)

230(+-10%)50Hz

Power consumption

VA

1100

3

Dry Helium Leak Detector UL5000

ACCESSORIESUL5000 Helium Leak Detector
Extension cable for remote control RC1000, 8 m length

14022

Helium Bottle Holder

551-001

LeakWare for UL1000/5000-Family

14090

Remote Control RC1000C, wired, with 4m coiled cable551-010

Remote Control RC1000WL, wireless, incl. wireless transmitter

551-015SL200, Sniffer Probe for UL1000/5000/Modul1000, 4m length

14005

Wireless transmitter for RC1000WL

551-020

www.inficon.com

reachus@inficon.com

Due to our continuing program of product improvements, specifications are subject to change without notice.
RateWatcher is a trademark of INFICON. All other trademarks are the property of their respective owners.
(2013-05)© 2013 INFICON