光波面計測システム

OEC

Optical Engineer
Companion

Metrology solution
The all-rounder
Metrology on-the-go
Modular system
Upgradablecompatible

光貿易株式会社 〒113-0034 東京都文京区湯島３－１３－８湯島不二ビル３０１
メール: contact@hikari-trading.com ウェブサイト: http://www.hikari-trading.com
TEL: 03-3832-3117 FAX: 03-3832-3118

OEC DATASHEET 2311

OPTICAL SETUPS
Concave surfaces / refractive
optics testing: use R-FLEXTM

Beam expanders
characterization and alignment(Uncoated) Small to large concave mirrors

Beam expanders, afocal systems
characterization and alignment

Flat wavefront / surfaces testing:use R-FLEXTM + R-FLEX LATM

Collimated lasers and external sources(Very) Large telescopes and collimators
characterization and alignment

Filters, windows in transmission

Lenses of any diameter, on-axis

Mirrors, filters in reflection

Lenses of any diameter, in the field

Head-Up displays, dichroics, beamsplitters

OEC DATASHEET 2311

光波面計測システム R-FLEXの構成
OPTICAL ENGINEER COMPANION
Our family of products
now teams up to support
your optical metrology
Compatible with the
Optical Engineer
Companion modularsystem: easily combine
the accessories youneed.

CONFIGURATIONThe OEC refers to a family of standard sensors and accessories thatcan be instantaneously combined to create the configuration you
need for your optical tests.
Select the modules among:a variety ofmetrology
sources from
UV to SWIRa large range
of HASOTM or
LIFT wavefront
sensors

several
R-FLEXTM
illumination
systemsa choice of
R-FLEX LATM
beam
expanders...and use them in combination or separately.

#1
測定用光源
METROLOGY SOURCE: MS-ʎ#2
光波面センサーWAVEFRONT SENSOR: HASOTM & LIFT

HASO LIFT 680

MS-ʎ

HASO LIFT 272

400 - 1550 nm

680 x 504 phase points

272 x 200 phase points

Perfect-wavefront metrology
source:+ Laser Diode or SLED+ Monomode fiber+ FC/APC output+ Manual or remote (USB) control+ Adjustable intensity output+ Trigger input+ Optical power: 5 mW

HASO4 126

HASO4 BROADBAND

Reference

350 - 1100 nm

HASO4 126 BROADBAND
13.8 x 10.2 mm

Wavelength (nm)MS-405405

MS-488488

MS-520520

MS-635635

MS-785785

MS-830830MS-1064

1064

MS-1550

1550

MS-CUSTOM

350 - 1100 nm

contact us for specificwavelength

HASO4 FIRSTHASO4 FAST

HASO SWIR LIFT

HASO SWIRl/100 RMS at custom wavelength

160 x 128 phase points

1.25 kHz

0.9 - 1.7 mm

OEC DATASHEET 2311

#3
R-FLEX、集光モジュール
ILLUMINATION SYSTEM: R-FLEXTM

R-FLEX2 VIS (BB)

400 - 750 nm (400 - 1100 nm)

R-FLEX2 SWIR

R-FLEX VIS and SWIR are
illumination systems coupling
our sources with our wavefront
sensors for testing:+ in double pass opticalconfiguration+ concave surfaces or refractive
optics of any f/# choosing theappropriate focusing module(MOD F) below:

1 - 1.7 mm

Reference
MOD F4.5
MOD F9MOD F9 HR
MOD F18 HRMOD F20
MOD F20 SWIR
MOD F31
MOD F31 SWIR
MOD F40
MOD F40 SWIR
MOD F50
MOD F50 SWIR
MOD F60
MOD F60 SWIR
MOD F75
MOD F76 SWIR
MOD CUSTOM

Compact, lightweight,
insensitive to vibrations
and atmosphericturbulences, R-FLEX iseasy to align onto theoptics to be tested.
It is the perfect solution forlarge telescopes, optics invacuum chambers.

HASO
HASO HASO4 126 /
HASO4
HASO4
LIFT 680 LIFT 272 BROADBAND BROADBAND FIRST

HASO4
HASO
FAST SWIR LIFT

HASO
SWIR

Focal Back power Working Module
333f/#, (wfe3) f/#, (wfe3) f/#, (wfe3) f/#, (wfe3) f/#, (wfe3)
length required distance1 length2 f/#, (wfe ) f/#, (wfe ) f/#, (wfe )(nmRMS)(nmRMS)(nmRMS)(nmRMS) (nm RMS) (nm RMS) (nm RMS) (nm RMS)
(mm)
(mm)
(mm)
(%)
4.5
50
0.6
205.9
0.9 (120)
0.9 (120)
1.3 (30)
3.8 (5)
8.950
1.2
250.0
1.8 (40)
1.8 (40)
2.5 (10)7.5 (5)950
0.6
247.8 0.9 (130)0.9 (130)17.8501.2
247.2 1.8 (50)
1.8 (50)
20.4
3
10.4
50.3
4 (160)4 (160)5.6 (40) 17.2 (5)
20
310
71.8
2.7 (250) 2.7 (250)
30.3
3-0.7
66.6
3 (250) 5.9 (20)
3 (250)5.9 (20)
8.4 (10) 25.5 (5)
31
3
10
114.7
4.2 (150) 4.2 (150)41.7
3840.8 4.1 (150) 8.1 (15)
4.1 (150)
8.1 (15)
11.5 (10) 35.1 (5)
40
3
3.3
62.2
5.4 (100) 5.4 (100)
51.7
3
3.6
103.1 5.1 (100) 10 (10)
5.1 (100)
10 (10)14.2 (5) 43.5 (5)50
3
3.3
92.16,8 (25) 6,8 (25)
60.1
3
15.2
42.5
5.9 (30) 11.7 (8)5.9 (30)
11.7 (8)16.5 (5) 50.5 (5)
60
3
12.8
45
8.1 (50) 8.1 (50)
75.1
3
12.8
73.6
7.4 (20) 14.6 (5)7.4 (20)
14.6 (5)20.6 (5) 63.1 (5)
75.3
3
12.8
74.3
10.2 (25) 10.2 (25)contact us for specific aperture, working distance, etc.

All focusing modules have pupil imaging: the microlens array of the wavefront sensor is imaged at the infinity by the focusing modules
1Distance between the focal plane and the first mechanical interface of the module (with centering tool removed)
2
Distance from the mounting interface of the module to the first mechanical interface of the module (with centering tool removed)
3
Wavefront Error (wfe) at the output of the module for a circular pupil corresponding to the nominal f-number (f/#)#4
ビームエキスパンダー（必要であれば）BEAM EXPANDER: R-FLEX LATM

R-FLEX LA VIS (BB)

400 - 750 nm (400 - 1100 nm)

R-FLEX LA VIS and SWIR are beam
expander coupling our R-FLEX for
testing:+ in double pass optical configurationor single pass for large beam lasers+ flat wavefront or surfaces of anydiameter choosing the appropriate
platform reference below:
Reference

R-FLEX LA SWIR
1 - 1.7 mm

Test diameter
Spectral range
(mm)

LA.30

30

VIS, SWIR

LA.50

50

VIS, SWIR

LA.75

75

VIS, SWIR

LA.100100

VIS, SWIR

LA.150150

VIS, SWIR

LA.200200

SWIR

At wavelength,
easy alignment andcharacterization ofbeam expanders
filters, dichroic beamsplitterflat mirrors
optical windows
polarization scramblers.

LA.CUSTOM contact us for specific diameter
OEC DATASHEET 2311