CP06E 【バイオチップ／CP06E】PepTenCamera＜FD-15＞(eng)

PepTenCam ＜PTC-FD15＞ ON SITE use
Design and Concept
HiPep Laboratories established fundamental*PC istechnologies for the PepTenChip®, system,not
peptide microarray, using a novel carbon
included
substrate realized the novel bio-detection
principle applied the fingerprint method (Pat.
US, EU, JP). Its ripple effect is very large, notonly research and drug discovery but also
involving high potential in clinical diagnoses
in t h e next generation. Bio-chips as films
Insertcameras are indispensable. Conventional
PepTenChip®
detection system using sophisticated with big
instruments with c o m p l i c a t e d r e a g e n t systems require expertized skill, which canbe performed only big institutions/hospitals.
Bio-detection using PepTenChip ® allows a“real-time + on-site ” manner, directly
measurements of specimen give time saving.
Thus a portable and easy-to-use detectiond e v i c e, “PepTenCam” has been developed
Specification: LED Excitation; Digital sCMOS camera; Image(PAT, JP & CN). “Easy to use in a remote
capture software; Easily carried, ca 6 Kg, carry on baggage size:
place/in a field” allows also for home care.
H 37 x D 25.3 x W13.2 cm, Inside power supply and USB-PC
Bio-safely level facility (BSL3, 4) is required
when the samples have infection risks. Inconnection. Easy installation & Operation. Maintenance freesuch closed space, PepTenCam is useful.
No sliding parts are equipped to allow maintenance free operation. PepTenCam had been manufactured for upperstream patens owned by HiPep Laboratories. Additionally this device is also useful for observation using micro-flow
path (on-chip manner) (US, JP and EU patents approved).
In bio-detection PepTenChip® can create objective indicators in diagnoses, not dependent on the skill of clinicians.
In care of serious diseases and/or senior persons who can not fully communicate their problems orally to doctors.

Additional filters: excitation wavelength for fluorescence (optional)
The wavelength of the excitation light source is optimized for TAMRA on capture
molecules. Light sources and filters can be customized upon requests.

Prototype
FD13

Wave length can be selected upon dyes and the filter can be easily replaced
TAMRA (Ex. 531 nm Em. 593 nm) ; FAM (Ex. 475 nm Em. 530 nm )Previousmodel
FD11Detection software (optional)
The software is offered as an option,
which is specialized in detecting andanalyzing arrayed spots.
By applying the optimal template to
the array image acquired by thePepTenCamera,
thedifference
between the fluorescence intensityof each spot and the backgroundcan be automatically calculated anddisplayed. The image data (arraydata) acquired by the otherequipment can be also analyzed.
This software was constructed as
outsourcing of HiPep Laboratories alicense fee is required. Practicallyanother software can also beapplied.

HiPep are looking for
manufactures underlicense

Previous prototype models, are offered
at a reduced price. Please contact us.
Hence, the specifications are thesame as FD15 (below with top-cover)except size.

SUMMARY➢ Portable➢ Digital sCMOS camera➢ Maintenance free➢ Image capture software➢ Easy handling➢ LED Excitation➢ Space saving➢ Filter selectable upon dyes

➢ On-site use➢ Bio-Safety-Level 3 or 4

Specifications will be changed without any notice

HiPep Laboratories, Nakatsukasa-cho 486-46, Kamigyo-ku, Kyoto, 602-8158, Japan
Tel +81-75-813-2101 E-mail: PepTenChip@hipep.jp URL: http://www.hipep.jp/eng