MR SERIES CARD LOGGERSSPLASH PROOF TYPE TEMPERATURE CARD LOGGER
MODEL MR5300
Model MR5300 is a small size andlightweight card logger capable to store up
to 6000 temperature data, and is washable(IP64 - splash proof).
Various types of external mounting
sensors including a built-in type areavailable.
Recalling of data and programming oflogging parameters can be executed withthis card logger or through a personal
computer.
This card logger conforms to CE and IP-64.
 FEATURES A traceability or inspection certificate, which canbe used for HACCP or ISO9000, is available.
(Separate purchase required) IP 64 (dust tight and splash proof) enclosure CE conformance High accuracy ±0.5ºC (for –5 to 50ºC)
 Automatic stop or endless of data logging isselectable.

 APPLICATIONS Small/lightweight and easy-to-operate Temperature monitoring in transportation

 Built-in totalizing function

 Freshness monitoring of vegetables or fruits

 Keeping of stored data for long term on lowbattery Temperature monitoring in show cases Temperature monitoring in refrigerators/freezers

 Transportation/storage temperature monitoringof pharmaceuticals

 Stored data can be recalled during logging ofdata.

 Temperature monitoring in green/vinyl houses
 Measurement interval of 10 to 50 seconds or 1
to 60 minutes

 Temperature monitoring in chambers

 Storing data can be read out through a readout
unit to PC (option)

 Temperaturemonitoring
trucks/warehouses

inrefrigerated

 Temperature monitoring in cement curing
1

PSE-481D SPECIFICATIONS (Model MR5300)
Measuring range:-40 to 60ºC, 0 to 100ºC, 50 to 150ºC (0 to 50ºC is available. For display only)(Ranges are automatically switched by judging a sensor to be used.)

Accuracy rating:±1.0ºC ± 1 digit (±0.5ºC ± 1 digit for –5 to 50ºC)

Sensor:

Built-in/External switching (Automatically switched to External when an external
mounting sensor is inserted. Measuring range of the built-in sensor: –40 to 60ºC)

Input point:One

Number of data stored:

6000 data (About 8-month data storage at the 60-minute measurement interval)

Storing mode:

Automatic stop mode under specified conditions or endless mode by a PC

Communications function:

Built-in (Connection to a PC through a data readout unit)

Display:

Reflection type liquid crystal display

Display contents:

Current data: Month, day, hour, minute, temperature (ºC)
Recalling data:
Data being stored per month, day, hour, minute or the following data in thestored data Totalized data (ºCH) of lower (L) or higher (H) temperature than a referencetemperature
 Number of activated alarms for higher or lower limit, and month, day, hour,
minute and its temperature of activated alarm Maximum, minimum and average temperatures
Parameters:
Clock: Year, month, day, hour, and minute (Corresponding to leap years)
Start time: Month, day, hour, and minute
Measurement interval: 1 to 60 minutes (1-minute increments) or 10 to 50
seconds (10-second increments)
Reference temperature: Reference temperature for totalization
Low battery: Display will blink when the batteries are low.

Totalization:

High or low totalization

Alarm:High or low alarm

Battery:

AAA (UM-4), 2 pieces
Battery life: About 1 year (At 25ºC and 10-minute interval)

Working environment:-20 to 55 ºC; 10 to 80%RH (No condensation)

Storage environment:-20 to 55ºC; under 90%RH (No condensation)

Housing structure:

IP64 (Splashing proof)

Material/color:

ABS Resin, PANTONE® color (equivalent to 552C)

Outside dimensions/weight: 54 (W) x 110 (H) x 20.5 (D) mm, 95g (Including batteries)
CE-marking:EN-61326, Emission: Class B, Immunity: Requirement for portable test andmeasurement equipment

 DIMENSIONS

 Convenience totalizing function
H totalized value (ºCH)= (Measured Temperature - Reference Temperature) x Hours
L totalized value (ºCH)= (Reference Temperature - Measured Temperature) x Hours*The totalizing function only executes when the value in() is over than 0ºC. The range of the totalizing
temperature is from 0 to 99990ºCH.
(Example)
Reference temperature: 2.0ºC
Measured temperature of 25.5ºC for 100 hours and5.5ºC for 50 hours
The H totalized value becomes 2525 (ºCH).
(25.5 - 2.0) x 100 + (5.5 - 2.0) x 50 =2525

Unit: mm2

For simultaneous data logging oftemperature in 2 places
SPLASH PROOF TYPE TWO-CHANNEL CARD LOGGER
MODEL MR5320
ModelMR5320

isasmallsizeandlightweight 2-channel card logger capable
to monitor temperature in two places andstore up to 6000 data per channel, and iswashable (IP64 - splash proof).
By connecting this unit to a personal
computer through the data readout unit
MR9503, all data can be displayed orprinted out in a table format or a trend
graph format, and also the data stored canbetreatedby worksheet

applicationscommercially available.
This card logger conforms to CE and IP-64.

 FEATURES
 Storing of 6000 data per channel

 Reliable infrared communications between thecard logger and the data readout unit without
any physical contact

 Measurement interval of 10 to 50 seconds or 1to 60 minutes

 IP 64 (dust tight and splash proof) enclosure High accuracy ±0.5ºC (for –5 to 50ºC)
 CE conformance
 Automatic stop or endless of data logging isselectable.

 Various kinds of sensors including a general
type, a water-proof type, a surface type and aneedle type

 Keeping of stored data for long term on lowbattery

 A traceability or inspection certificate, which canbe used for HACCP or ISO9000, is available.
(Separate purchase required) Storing of about 730 thousand data in a 1.44MB
floppy disk (by exclusive file format)

 APPLICATIONS Temperaturelaboratories

monitoringinresearch Monitoring of multiple temperature in various
kinds of applications

and Monitoring of temperature difference Temperature monitoring in green/vinyl houses3 SPECIFICATIONS (Model MR5320)
Measuring range:-40 to 60ºC, 0 to 100ºC, 50 to 150ºC(Ranges are automatically switched by judging a sensor to be used.)* With the built-in sensor (Channel 1 only): -40 to 60ºC

Accuracy rating:±0.5ºC ± 1 digit (for –5 to 50ºC)* Except above: ±1.0ºC ± 1 digitSensor:

Built-in (Channel 1 only)/External switching* Various types of sensors (separate purchase required) including general type,water-proof type, needle type and surface type are available.

Input point:

Two points

Number of data stored:

6000 data/each channel

Display contents:(Each channel)

Current data:
Month, day, hour, minute, measured temperature in Channel 1 and Channel 2
Recalling data:
Maximum temperature, minimum temperature and average temperature instored data in Channel 1 and 2Parameters:
High alarm set point and low alarm set point of Channel 1 and 2

Operation by logger:

Display start-up (With auto-off function: Display goes out after one minute.)
Display content selection (clock setting, logging start/stop, key lock/release)* Three function keys of DISP, SEL and REC

Communications function:

Infrared communications to the exclusive data readout unit (separate purchaserequired)
Settings of parameters and recalling of stored data/logging parameters by theexclusive analytical software package

Settings by PC:(Exclusive analytical software)

Clock, logging start time/stop time (in case of the measurement interval of 1 to 60minutes), measurement/logging interval (10 to 50 seconds or 1 to 60 minutes),high/low alarm temperatures, key locking of logger, logging mode (automaticstop/endless)Battery:AAA (UM-4), 2 pieces

Battery life:

About 1 year (At 25ºC and 1-minute interval)

Outside dimensions/weight: 54 (W) x 110 (H) x 20.5 (D) mm, 95g (Including batteries)
CE-marking:EN-61326, Emission: Class B, Immunity: Requirement for portable test andmeasurement equipment

 MOUNTING HOLDER DIMENSIONS(Common to MR5300, MR5320 and MR662)
This is used to fix the card logger to equipment, ironpoles or wall.
Model: MR9013
Mounting: By screws
Material: SUS304

Unit: mm
Unit: mm4

 EXTERNAL MOUNTING TEMPERATURE SENSORS (common to MR5300 and MR5320)
Unit: mm

 General type
MR9301 (-40 to 60ºC)
MR9302 (0 to 100ºC)
MR9303 (50 to 150ºC)

 With long lead wire

 With water-proof tube
MR9301S30 (-40 to 60ºC)
MR9302S30 (0 to 100ºC)

 With metal tube
MR9301A30 (-40 to 60ºC)
MR9302A30 (0 to 100ºC)
MR9303A30 (50 to 150ºC)

 With metal tube (sanitary)
MR9301B30 (-40 to 60ºC)
MR9302B30 (0 to 100ºC)
MR9303B30 (50 to 150ºC)

 Surface temperature
MR9301F30 (-40 to 60ºC)
MR9302F30 (0 to 100ºC)

MR9301M05 (-40 to 60ºC/5m)
MR9302M05 (0 to 100ºC/5m))
MR9303M05 (50 to 150ºC/5m)
MR9301M10 (-40 to 60ºC/10m)
MR9302M10 (0 to 100ºC/10m))
MR9303M10 (50 to 150ºC/10m)
MR9301M15 (-40 to 60ºC/15m)
MR9302M15 (0 to 100ºC/15m))
MR9303M15 (50 to 150ºC/15m)

 Needle type (with finger holder)MR9301D05 (-40 to 60ºC/sensor 50mm)
MR9302D05 (0 to 100ºC/ sensor 50mm)
MR9301D10 (-40 to 60ºC/sensor 100mm)
MR9302D10 (0 to 100ºC/ sensor 100mm)
MR9301D20 (-40 to 60ºC/sensor 200mm)
MR9302D20 (0 to 100ºC/ sensor 200mm)

Remarks:
For CE conformance, the external mounting sensors of Models 9301, 9302 and 9303 only areapplicable to MR5300 or MR5320.
5

TEMPERATURE HUMIDITY CARD LOGGER
MODEL MR6662
The MR6662 small size and lightweightcard logger is for simultaneous logging oftemperature and humidity and is capable
to store up to 6000 data (40days data at
10-minute measurement interval) oftemperature and humidity. A large screen
enables to display the data being stored atsite. Software for data management
through a personal computer is separately
offered with a data readout unit. For
humidity measurement, aninterchangeable element of capacitancetype high polymer film is used. This cardlogger conforms to CE.

 FEATURES
 Simultaneous measurement and logging oftemperature and humidity
 Up to 6000 data storage of temperature and APPLICATIONS

humidity Temperature and humidity control in computer Interchangeable humidity element (capacitancerooms

type high polymer film)

 Temperature and humidity control in plants of

 Programmable measurement interval from 1 toelectronic equipment

60 minutes (At the measurement interval of 10

 Inspected data recording of housing-related

minutes, data are stored up to about 40 days.)instruments Small/lightweight and easy-to-operate Temperature and humidity control in museums

 Recalling of stored data and settings of logging

 Freshness control of vegetables, flowers, etc.

parameters by this logger or through a personal Temperature and humidity control in constantcomputertemperature warehouses and containers Keeping of stored data by backup function on Environmental

low battery

controlofelectronicandcommunication equipment

 A traceability or inspection certificate, which can Measurement of temperature and humidity in

be used for HACCP or ISO9000, is available.

green houses(Separate purchase required) Environmental measurement of temperature CE conformanceand humidity
 For teaching tools in schools6 SPECIFICATIONS (Model MR6662)
Measuring range:

Temperature
Humidity-40 to 60ºC
0 to 100% (no dew condensation)

Accuracy rating:Temperature±0.5ºC (for –5 to 50ºC)(Except above: ±1.0ºC)±3%RH (at 25ºC ± 2ºC/0 to 90%RH)

Humidity
Input point:

Temperature 1 point, humidity 1 point

Number of data stored:

Temperature/humidity each 6000 data (Data storing for about 40 days at themeasurement interval of 10 minutes)

Display contents:

Current data:
Month, day, hour, minute, temperature (ºC), humidity (%RH)
Recalling data:
Data being stored per month, day, hour or minuteMaximum temperature/humidity, minimum temperature/humidity and averagetemperature/humidity
Parameters:
Clock – Year, month, day, hour, minute (corresponding to leap years)
Start time - Year, month, day, hour, minute
Logging interval – 1 to 60 minutes (1-minute increments)
High/low alarm set points

Alarm:

ALARM appears on alarm activation.

Working environment:

Logger - -20 to 55ºC
Temperature/humidity sensor - -30 to 60ºC (no dew condensation)

Outside dimensions/weight: Logger - 54 (W) x 110 (H) x 20.4 (D) mm
Sensor - 18 (W) x 53 (H) x 11 (D) mm
Weight:95g (including batteries)

Battery:

AAA (UM-4), 2 pieces

Battery life:

About 1 year (At 25ºC and 10-minute interval)

CE-marking:EN-55011, Group 1, Class B, EN-50082-1

 MODEL

NameMain Unit
Temp/ Humidity
Cable (1m)Model
MR6662
MR9202
MR9282-010

 MOUNTING HOLDER DIMENSIONS(Common to MR5300, MR5320 and MR6662)
This is used to fix the card logger to equipment, ironpoles or wall.
Model: MR9013
Mounting: By screws
Material: SUS304

Unit: mm

Unit: mm7

DATA READOUT UNIT (applicable to all card loggers)
MODEL MR9504
Card loggers can display or printed out data in a table format or a trend graph format byconnecting to a personal computer through this data readout unit. The data stored canbe managed by a file and treated by worksheet applications commercially available. The
card loggers can connect to this data readout unit through reliable infraredcommunications without any physical contact.
 SPECIFICATIONS (With analytical software package)Model
USB connector
Environment
Communications
Weight

MR9504
Type A
1.4m
Windows XP sp3 or later/ Vista (32bit) sp 2 or later/ Win 7 (32bit/64bit)/ Win 8/8.1 (32bit/64bit)/ Win 10 (32 bit/64bit)
Hard disk: 256MB free space or more
Between a card logger and this data readout unit: Non-contact infrared type (4800 bps)
Between this data readout unit and a PC: USB ver. 2.0
About 180g

 ANALYTICAL SOFTWARE PACKAGE
ModelSettings
Recalling
Computation
Display
Data storage
Printout

MR5300MR5320MR6662Clock, start time, measuring interval, totalizing direction, card No., test No., Key lock/release (MR5320 only)
Logging type, direction of Logging type, direction ofLogging type, direction ofintegration, high/low alarm set integration, high/low alarm setintegration, reference value
pointspoints, reference value
Parameters, stored data, number of errors
Maximum value, minimum value, average value
Totalized value---Totalized value
All data (measured data per year, month, day, hour, or minute)
Trend graph---High/low alarms (alarm mark attached to the data)
Storage in file format or CSV, Text format by this software package
List printing, parameter printing, trend graph DIMENSIONS

Unit: mm

Specifications subject to change without notice. Printed in Japan (I)32-8, KUMANO-CHO, ITABASHI-KU, TOKYO 173-8632
PHONE: +81-3-3956-2171
FAX: +81-3-3956-0915
E-mail: inter@chino.co.jp
Website: http://www.chino.co.jp

82020.11