SNA6000A Series
Vector Network
Analyzer

Datasheet
DS09050_E01A

0

2 Features

SNA6000A

Frequency range: 100 kHz - 13.5 GHz and
100 kHz - 26.5 GHz
Frequency resolution: 1 Hz
Level resolution: 0.05 dB
Range of IFBW: 1 Hz~10 MHz

1 General Description

Setting range of output level:

The SIGLENT SNA6000A series of Vector
Network Analyzers have a frequency rangeof 100 kHz to 13.5 GHz and 100 kHz to 26.5
GHz, which support 2/4-port scatteringparameter, differential-parameter, and timedomain

parameter

measurements.

The

SNA6000A series of VNAs are effective
instrumentation for determining the Q-factor,-55 dBm ~ +10 dBm
Dynamic range: 135 dB
Types of calibration: Response calibration,
Enhanced Response calibration, Full-one
port calibration, Full-two port calibration,
Full-three port calibration, Full-four port
calibration, TRL calibration

bandwidth, and insertion loss of a filter, They

Types of measurement: Scattering-

feature impedance conversion, movement ofparameter measurement, differential-

measurement plane, limit testing, ripple test,

parameter measurement, receiver

fixturemeasurement, time-domain parameter

simulation,andadapterremoval/insertion adjustments. The VNAs

analysis, limit test, ripple test, impedance

have five sweep types: Linear-Frequency

conversion, fixture simulation, adapter

mode, Log-Frequency mode, Power-Sweep

removal/insertion, spectrum analysis

mode, CW-Time mode, and Segment-Sweep

frequency offset, scalar mixer

mode. The SNA6000A series VNAs alsomeasurement, pulse measurement,

support scattering-parameter correction ofMaterial Measurement

SOLT,

SOLR,

TRL,

Response,andEnhanced Response for increased flexibility
in R&D and manufacturing applications.

Internal Bias-Tee connections
Interface: LAN, USB Device, USB Host
(USB-GPIB)
Remote control: SCPI/ Labview/ IVI basedon USB-TMC / VXI-11 / Socket /Telnet /WebServer
12.1-inch touch screen
Video output: HDMI/DVI-D/DP

SNA6000A DataSheet

13 Models and key specifications
Model

SNA6034A

SNA6032A

SNA6024A

SNA6022A

SNA6134A

SNA6132A

SNA6124A

SNA6122A

Frequency range

100kHz-26.5GHz

Ports

4

Frequency resolution

1 Hz

Level resolution

0.05 dB

Range of IFBW

1 Hz~10 MHz

Number of points

2 to 100,001

Setting range of
output level2

100kHz-13.5GHz
4

2-55 dBm ~ +10 dBm

Dynamic range

135 dB

Types of calibration

Response calibration, Enhanced Response calibration, Full-one port
calibration, Full-two port calibration, Full-three port calibration, Full-four
port calibration, TRL calibration

Types of

Scattering-parameter measurement, differential-parametermeasurement

measurement, receiver measurement, time-domain parameter analysis,
limit test, ripple test, impedance conversion, fixture simulation, adapter
removal/insertion, enhanced time-domain parameter analysis (TDR),
spectrum analysis, frequency offset, scalar mixer measurement, pulse
measurement, Material Measurement

Bias-Tees

SupportInterfaceLAN, USB Device, USB Host(USB-GPIB)

Remote control

SCPI/ Labview/ IVI based on USB-TMC/ VXI-11/ Socket/ Telnet/
WebServer

Display

12.1-inch touch screen

Video output

HDMI/DVI-D/DP

SNA6000A DataSheet

14 Design Features
Multi-window display:

Multi-format display:2

SNA6000A DataSheetDisplay and compare memory and current data:

Display data hold:

SNA6000A DataSheet3

Impedance conversion:

Equation Editor:

4

SNA6000A DataSheetPort Extensions:Embedding and De-Embedding:

SNA6000A DataSheet

5

Time-Domain analysis

Enhanced Time-Domain analysis(TDR)

6

SNA6000A DataSheet

Spectrum analysis

Scalar mixer measurement

SNA6000A DataSheet

7

Pulse Modulation

Material Measurement

8

SNA6000A DataSheetSwitch matrix measurement

SNA6000A DataSheet

95 Definitions
Specifications are valid under the following conditions: The instrument is within the calibration period,has been stored between 0 and 40℃ for at least 2 hours before use, and has been powered on andwarmed up for at least 90 minutes. The specifications include the measurement uncertainty unlessotherwise noted.
Specification: All products are guaranteed to meet published specifications at room temperature(approximately 25℃), unless otherwise noted.
Typical: Performance deemed typical implies that 80 percent of the measurement results will meet
the typical published performance with a 95th percentile confidence level at room temperature(approximately 25℃). Typical performance is not warranted and does not include measurement
uncertainty.
Nominal: This value indicates the expected mean or average performance, or an attribute whose
performance is by design, such as the 50 Ohm connector.

6 Specifications
6.1

Dynamic range

Frequency range

Specification(dB)

Typical (dB)

100 kHz- 1 MHz

120

125

1 MHz- 500 MHz

125

135

130

139

135

142

20 GHz- 24 GHz

127

135

24 GHz- 26.5 GHz

120

128

500 MHz- 1 GHz
1 GHz- 20 GHz

10

IFBW

10Hz

S N A 6 0 0 0 A D a ta S h e e t

6.2

Corrected system performance with calibration kit

User correction: On, system correction: On; Corrected system performance with Keysight 85052D
3.5mm calibration kit, isolation calibration performed.IFBW is 10 Hz, no averaging applied to data,and environmental temperature is 25℃ (± 5℃), with < 1℃ deviation from calibration temperature.
Specification (dB)

100 kHz-9 GHz

9 GHz-20 GHz

20 GHz-26.5 GHz

Directivity

41

36

35

Source match

36

29

27

Load match

41

36

33

Reflect tracking±0.004±0.003±0.01

Transmission tracking

±0.06

±0.09

±0.5

Reflection uncertainty (Specification, Power: -10 dBm, IFBW:10 Hz):

Transmission uncertainty (Specification, Power: -10 dBm, IFBW:10 Hz):

SNA6000A DataSheet

11

6.3

Uncorrected system performance

User correction: Off, system correction: On; IFBW is 10 Hz, no averaging applied to data.
Specification (dB)

100 KHz-1 GHz

1 GHz-9 GHz

9 GHz-20 GHz

20 GHz-26.5 GHz

Directivity

20

20

16

13

Source match

20

20

16

13

Load match

8

11

7

7

Reflect tracking

±1.4

±1.4

±1

±1

Transmission tracking

±1.4

±1.4

±1

±1

6.4

Test port output (Source)

6.4.1

Test port output frequency

Description

Specification

Frequency range
SNA6034A/ SNA6032A/
SNA6134A/ SNA6132A
SNA6024A/ SNA6022A/
SNA6124A/ SNA6122A
Frequency resolution

100 kHz to 26.5 GHz

100 kHz to 13.5 GHz
1 Hz

CW accuracyStandard± 1.0 ppm (23 ± 3℃ )Option: SNA6000-HPR± 0.1 ppm (23 ± 3℃ )Source stability
Standard± 1.0 ppm (0 to 40℃ )± 0.5 ppm/year，± 3.0 ppm/20 year

Option: SNA6000-HPR± 1 ppb (0 to 40℃ )± 50 ppb/year

12

S N A 6 0 0 0 A D a ta S h e e t

6.4.2

Test port output power
Description

Preset power

Specification

Typical

0 dBm

Level accuracy
100 kHz - 10 MHz±2.0 dB@0 dBm

10 MHz - 20 GHz±1.5 dB@0 dBm

20 GHz- 26.5 GHz±2.0 dB@0 dBm

Level linearity
100 kHz- 1 MHz±0.5 dB（-20 dBm to 10 dBm）

1 MHz- 500 MHz±0.5 dB（-20 dBm to 10 dBm）

500 MHz- 1 GHz±0.5 dB（-20 dBm to 10 dBm）

1 GHz- 20 GHz±0.5 dB（-20 dBm to 10 dBm）

20 GHz- 24 GHz±0.5 dB（-20 dBm to 7 dBm）

24 GHz- 26.5 GHz±0.5 dB（-20 dBm to 5 dBm）

Range
100 kHz- 1 MHz-55 dBm to 10 dBm

1 MHz- 500 MHz-55 dBm to 10 dBm

500 MHz- 1 GHz-55 dBm to 10 dBm

1 GHz- 20 GHz-55 dBm to 10 dBm

20 GHz- 24 GHz-55 dBm to 7 dBm

24 GHz- 26.5 GHz-55 dBm to 5 dBm

Sweep range
100 kHz- 1 MHz-55 dBm to 10 dBm

1 MHz- 500 MHz-55 dBm to 10 dBm

500 MHz- 1 GHz-55 dBm to 10 dBm

1 GHz- 20 GHz-55 dBm to 10 dBm

20 GHz- 24 GHz-55 dBm to 7 dBm

24 GHz- 26.5 GHz-55 dBm to 5 dBm

Max leveled power
100 kHz- 1 MHz

10 dBm

11 dBm1 MHz- 500 MHz

10 dBm

13 dBm

500 MHz- 1 GHz

10 dBm

12 dBm1 GHz- 20 GHz

10 dBm

10 dBmS N A 6 0 0 0 A D a t a Sh e e t

13

20 GHz- 24 GHz

7 dBm

10 dBm24 GHz- 26.5 GHz

5 dBm

8 dBm
0.05 dB

Level resolution

6.4.3

Test port output signal purity

Description

Specification

Typical2nd or 3rd harmonics (0 dBm)
100 kHz to 10 MHz<–20 dBc

10 MHz to 26.5 GHz<–25 dBc

Non-harmonic spurious (0 dBm)<-30 dBc

14

S N A 6 0 0 0 A D a ta S h e e t

6.5

Test port input

6.5.1

Test port input levels

Description

Specification

Typical

Max input level
100 kHz-26.5 GHz+10 dBm

Damage input level
100 kHz-26.5 GHz+27 dBm (RF) or 35 V (DC)Level accuracy
100 kHz - 10 MHz±2.5 dB@0 dBm

10 MHz - 20 GHz±1.5 dB@0 dBm

20 GHz- 26.5 GHz±2.0 dB@0 dBm

Crosstalk
100 kHz- 500 kHz-95 dB-110 dB

500 kHz- 5 MHz-110 dB-140 dB

5 MHz- 13.5 GHz-120 dB-125 dB

13.5 GHz- 26.5 GHz-108 dB-115 dB

100 kHz- 500 MHz-115 dBm/Hz-125 dBm/Hz

500 MHz- 1 GHz-125 dBm/Hz-135 dBm/Hz

1 GHz- 8 GHz-130 dBm/Hz-135 dBm/Hz

8 GHz- 20 GHz-135 dBm/Hz-135 dBm/Hz

20 GHz- 26.5 GHz-130 dBm/Hz-135 dBm/Hz

100 kHz- 500 MHz-115 dBm/Hz-130 dBm/Hz

100 kHz- 13.5 GHz

0.5 dB

0.3 dB

13.5 GHz- 26.5 GHz

1.0 dB

0.5 dB

100 kHz- 13.5 GHz

5 deg

1 deg

13.5 GHz- 26.5 GHz

5 deg

1.5 deg

Noise floor

Compression level(+10 dBm)
MagnitudePhase

S N A 6 0 0 0 A D a t a Sh e e t

15

6.5.2

Trace noise

Description

Specification

Typical

100 kHz- 10 MHz (IFBW=1 kHz)0.005 dB rms

0.0015 dB rms

10 MHz- 13.5 GHz (IFBW=10 kHz)0.009 dB rms

0.0015 dB rms

13.5 GHz- 26.5 GHz (IFBW=10 kHz)

0.015 dB rms

0.0025 dB rms

100 kHz- 10 MHz (IFBW=1 kHz)0.005 dB rms

0.0015 dB rms

10 MHz- 13.5 GHz (IFBW=10 kHz)0.009 dB rms

0.0015 dB rms

13.5 GHz- 26.5 GHz (IFBW=10 kHz)

0.012 dB rms

0.0025 dB rms

100 kHz- 10 MHz (IFBW=1 kHz)

0.012 deg rms

0.02 deg rms

10 MHz- 13.5 GHz (IFBW=10 kHz)

0.05 deg rms

0.015 deg rms

13.5 GHz- 26.5 GHz (IFBW=10 kHz)

0.05 deg rms

0.02 deg rms

100 kHz- 10 MHz (IFBW=1 kHz)

0.01 deg rms

0.015 deg rms

10 MHz- 13.5 GHz (IFBW=10 kHz)

0.05 deg rms

0.015 deg rms

13.5 GHz- 26.5 GHz (IFBW=10 kHz)

0.05 deg rms

0.02 deg rms

Note:Setting max output power
Transmission trace noise magnitude

Reflection trace noise magnitude

Transmission trace noise phase

Reflection trace noise phase

6.5.3

Stability

Description

Specification

TypicalMagnitude
100 kHz- 13.5 GHz± 0.01 dB/°C

13.5 GHz- 26.5 GHz± 0.05 dB/°C

Phase
100 kHz- 13.5 GHz± 0.1 deg/°C

13.5 GHz- 26.5 GHz± 0.9 deg/°C

16

S N A 6 0 0 0 A D a ta S h e e t

6.5.4

Dynamic accuracy

Description

Specification

Relative to -10 dBm input power
Magnitude
10 dBm

± 0.19 dB-30 dBm± 0.05 dB-100 dBm± 2.5 dB

Phase
10 dBm

± 4.5 deg

-30 dBm± 0.25 deg-100 dBm± 16.5 deg

S N A 6 0 0 0 A D a t a Sh e e t

17

6.6

Pulsed-RF

6.6.1

Pulse Modulation On/Off Ratio (dB)Description

Typical

100 kHz- 13.5 GHz

80

13.5 GHz- 26.5 GHz

70

6.6.2

Pulse Modulation

Description

TypicalMinimum pulse width

10 usec

Minimum pulse period

30 usec

Maximum pulse period

26 secPulse Modulation Shape Example

18

S N A 6 0 0 0 A D a ta S h e e t

7 Sweep time
Start frequency: 100 kHz, Stop frequency: 26.5GHz; IFBW: 500 kHz.
Points

201

401

1601

6401

Uncorrected

15 ms

17 ms

35 ms

141 ms2-port cal

30 ms

34 ms

70 ms

282 msStart frequency: 100 kHz, Stop frequency: 26.5GHz; IFBW: 100 kHz.
Points

201

401

1601

6401

Uncorrected

17 ms

20 ms

46 ms

185 ms2-port cal

34 ms

40 ms

92 ms

370 msStart frequency: 100 kHz, Stop frequency: 26.5GHz; IFBW: 10 kHz.
Points

201

401

1601

6401

Uncorrected

33 ms

52 ms

175 ms698 ms2-port cal

66 ms

104 ms

350 ms1396 ms

Start frequency: 100 kHz, Stop frequency: 26.5GHz; IFBW: 1 kHz.
Points

201

401

1601

6401

Uncorrected

193 ms372 ms1452 ms

5806 ms

2-port cal

386 ms

744 ms2904 ms

11612 ms

S N A 6 0 0 0 A D a t a Sh e e t

19

8 Enhanced Time Domain Analysis with TDR (SNA6000-TDR)
Description

SNA6022/4A、SNA6122/4A

SNA6032/4A、SNA6132/4A

Bandwidth13.5 GHz

26.5 GHz

Input Impedance

50 OhmDC damage Level

35 V

at test port
Maximum test port input

1.5Vppvoltage (Hot TDR Mode)
TDR stimulus

Step, Impulse

TDR step amplitude

1 mV to 5 V

TDR step rise time(min) (10% to 90%)

33.1 ps

16.9 ps

5 mm

2.5 mm44.7 ps

22.8 ps

TDR step response
resolution in free space(min) (εr = 1)
TDR impulse width (min)DUT length (max)
Eye diagram data rate (max)20

1.25 μs10.8 Gb/s

21.2 Gb/s

S N A 6 0 0 0 A D a ta S h e e t

9 General information
Description

Characteristics

Operating environment
Temperature

0 to 40 ℃

HumidityType tested at 20 to 80%, wet bulb temperature< 29 °C (non-condensing)

Altitude

0 to 3000 m

Non-operating storage environment
Temperature-20 ℃ to 60 ℃

HumidityType tested at 20 to 90%, wet bulb temperature< 40 °C (non-condensing)

Altitude

0 to 15000 m

Size

W×H×D=426×251×494.5 mm

Weight4-port or SNA6034A/SNA6134A: 19 kg

EMC
Conducted disturbance: CISPR 11/EN 55011

CLASS A group 1, 150 kHz - 30 MHz

Radiated disturbance: CISPR 11/EN 55011

CLASS A group 1, 30 MHz -1 GHz

Electrostatic discharge(ESD):

4.0 kV (contact), 8.0 kV (air)IEC61000-4-2/EN61000-4-2
Radio-frequency electromagnetic field Immunity:

10 V/m (80 MHz to 1 GHz);

IEC 61000-4-3/EN 61000-4-3

3 V/m (1.4 GHz to 2 GHz);
1 V/m (2.0 GHz to 2.7 GHz)Electrical fast transients (EFT):

2 kV (AC power ports)

IEC 61000-4-4/EN 61000-4-4
Surges: IEC 61000-4-5/EN 61000-4-5

1 kV (Line to line) ; 2 kV (Line to ground)Radio-frequency continuous conducted

3 V, 0.15-80 MHz

Immunity: IEC 61000-4-6/EN 61000-4-6
Voltage dips and interruptions:

Voltage dips:

IEC 61000-4-11/EN 61000-4-11

0% UT during 1 cycle;
40% UT during 10/12 cycles;
70% UT during 25/30 cycles;
Voltage interruptions: 0% UT during 250 cycles

SafetyUL 61010-1:2012/R: 2018-11; CAN/CSA-C22.2 No. 61010-1:2012/A1:2018-11.
UL 61010-2-030:2018; CAN/CSA-C22.2 No. 61010-2-030:2018.
S N A 6 0 0 0 A D a t a Sh e e t

21

10 Front panel information
Description

Characteristics

RF connectors

3.5mm NMD（male），50Ω（SNA6132/4A、SNA6122/4A）

Damage level+27 dBm or ±35 VDC

Display Resolution

12.1 inch TFT color LCD with touch screen ; WXGA (1280 x 800)

USB interface

USB-A 2.0

11 Rear panel information
Description

Characteristics

Ext trigger input connector
Type

BNC, female

Input level

5V TTLExt trigger output connector
Type

BNC, female

Max output current

20 mA

Output level

3.3V TTL

Ext ref-signal input connector
Type

BNC, female

Input frequency

10 MHz ±10 ppm

Input level-3 dBm to +10 dBm

Input impedance50Ω

Int ref-signal output connector
Type

BNC, female

Output frequency

10 MHz ± 1 ppm

Signal type

Sinewave

Output level

0 dBm ± 3 dB into 50 Ω

Output impedance

50 Ω

Bias tee input connector
Type

BNC, female

Max voltage± 35 VDC

Max current (no degradation RF specification)± 300 mA

Max current (damage level)

500 mA

Video output

HDMI/DVI-D/DP

USB (USBTMC) interface

USB-B 3.0, USB-A 3.0

LAN

10/100/1000 BaseT Ethernet

Power100~240 Vrms 50/60 Hz

Power consumption

4-port: 170 W (typical)

22

S N A 6 0 0 0 A D a ta S h e e t

12 Ordering Information
Items

Description

Order number

4 ports, 26.5G Vector Network Analyzer

SNA6034A

2 ports, 26.5G Vector Network Analyzer

SNA6032A

4 ports, 13.5G Vector Network Analyzer

SNA6024A

2 ports, 13.5G Vector Network Analyzer

SNA6022A

4 ports, 26.5G Vector Network Analyzer
Products

SNA6134A(Includes front panel jumper interface)
2 ports, 26.5G Vector Network Analyzer

SNA6132A(Includes front panel jumper interface)
4 ports, 13.5G Vector Network Analyzer

SNA6124A(Includes front panel jumper interface)
2 ports, 13.5G Vector Network Analyzer

SNA6122A(Includes front panel jumper interface)
Standard

1 x Quick-start, 1 x Power-cable, 1 x USB-cable, 1 x calibration-certificate,

Accessories

1 x Wireless mouse, 1 x Protective Cover

Optional
Accessories

High-performance reference source

SNA6000-HPR

Time-Domain analysis

SNA6000-TDA

Enhanced Time-Domain analysis

SNA6000-TDR

Spectrum analysis

SNA6000-SA

Scalar mixer measurement

SNA6000-SMM

Pulse measurement

SNA6000-PM

Material Measurement

SNA6000-MT

SEM5000A series electronic calibrators

SEM5000A

N-type, Male, 50Ω Calibration Kit, 0-4.5GHz

F503ME

N-type, Female, 50Ω Calibration Kit, 0-4.5GHz

F503FE

N-type, Male, 50Ω Calibration Kit, 0-9GHz

F504MS

N-type, Female, 50Ω Calibration Kit, 0-9GHz

F504FS

N-type, Male, 50Ω Calibration Kit, 0-9GHz

Y504MS

N-type, Female, 50Ω Calibration Kit, 0-9GHz

Y504FS

N-type, Male and Female, 50Ω Calibration Kit,0-9GHz

F504TS

N-type, Male and Female, 50Ω Calibration Kit,0-18GHz

F505TS

3.5 mm, Male, 50Ω Calibration Kit, 0-4.5GHz

F603MES N A 6 0 0 0 A D a t a Sh e e t

23

3.5 mm, Female, 50Ω Calibration Kit, 0-4.5GHz

F603FE

3.5 mm, Male, 50Ω Calibration Kit, 0-9GH

F604MS

3.5 mm, Female, 50Ω Calibration Kit, 0-9GHz

F604FS

3.5 mm, Male and Female, 50Ω Calibration Kit, 0-9GHz

F604TS

3.5 mm, Male, 50Ω Calibration Kit, 0-26.5GHz

Y606MS

3.5 mm, Female, 50Ω Calibration Kit, 0-26.5GHz

Y606FS

3.5 mm, Female, 50Ω Calibration Kit, 0-26.5GHz

F606FS

3.5 mm, Male and Female, 50Ω Calibration Kit, 0-26.5GHz

F606TS

50Ω Waveguide calibration kit, 18-26.5GHz

KWR42A

N(M)-SMA(F) RF Cable DC~6 GHz,1000 mm

S06-NMSF-1M

N(M)-SMA(F) RF Cable DC~18 GHz,1000 mm

S18-NMSF-1M

2.9 mm(M)- 2.9 mm (F) RF Cable DC~40 GHz,1000 mm

S40-29M29F-1M

N(M)-SMA(M) RF Cable DC~18 GHz,1000 mm

N-SMA-18L

N(M)-N(M) RF Cable DC~18 GHz,1000 mm

N-N-18L

SMA(M)-SMA(M) RF Cable DC~18 GHz,1000 mm

SMA-SMA-18L

SMA(M)-SMA(M) RF Cable DC~26.5 GHz,1000 mm

SMA-SMA-26L

SMA(F)-SMA(M) RF Cable DC~26.5 GHz,1000 mm

SMAF-SMA-26L

NMD 3.5 female-NMD 3.5 Male DC-26.5 GHz,635 mm

NMD 3.5 female-APC 3.5 female DC-26.5 GHz,635 mm

24

V26-N35MN35F25IN
V26-N35FA35F25IN

USB-GPIB Adapter

USB-GPIB

RF demonstration board

SNA-TB01

Adjustable Differential TDR probe DC-18 GHz

ADP-18Adjustable Differential TDR probe DC-26.5 GHz

ADP-26

Adjustable Single-end TDR probe DC-18 GHz

ASP-18Adjustable Single-end TDR probe DC-26.5 GHz

ASP-26

S N A 6 0 0 0 A D a ta S h e e tAbout SIGLENTSIGLENT is an international high-tech company, concentrating on R&D, sales,
production and services of electronic test & measurement instruments.

SIGLENT first began developing digital oscilloscopes independently in 2002.
After more than a decade of continuous development, SIGLENT has extended
its product line to include digital oscilloscopes, isolated handheld
oscilloscopes,

function/arbitrary

waveform

generators,

RF/MWsignalgenerators, spectrum analyzers, vector network analyzers, digital multimeters,
DC power supplies, electronic loads and other general purpose test
instrumentation. Since its first oscilloscope was launched in 2005, SIGLENT
has become the fastest growing manufacturer of digital oscilloscopes. We
firmly believe that today SIGLENT is the best value in electronic test &measurement.

Headquarters:
SIGLENT Technologies Co., Ltd
Add: Bldg No.4 & No.5, Antongda Industrial
Zone, 3rd Liuxian Road, Bao'an District,
Shenzhen, 518101, China

Tel: + 86 755 3688 7876
Fax: + 86 755 3359 1582
Email: sales@siglent.com
Website: int.siglent.com

North America:
SIGLENT Technologies America, Inc
6557 Cochran Rd Solon, Ohio 44139
Tel: 440-398-5800
Toll Free: 877-515-5551
Fax: 440-399-1211
Email: info@siglentna.com

Website: www.siglentna.com

Europe:SIGLENT Technologies Germany GmbH
Add: Staetzlinger Str. 70
86165 Augsburg, Germany
Tel: +49(0)-821-666 0 111 0
Fax: +49(0)-821-666 0 111 22
Email: info-eu@siglent.com
Website: www.siglenteu.com

Follow us on
Facebook: SiglentTech