SCOUT200* series
Portable vibration analyzers

These portable data collector solutions record the vibrationof industrial machines, to monitor their condition anddiagnose their faults. The Series includes “SCOUT” models
for use in hazardous areas (explosive atmospheres) and“COMMTEST” models for less stressful environments. Thereare 2-channel and 4-channel versions of each, to suit yourdata acquisition and analysis needs.

Designed in and for the21st century

These two-part solutions utilize the power andconnectivity of modern industrial smartphones
and tablets, paired with a small form factor,
rugged data acquisition module. This gives thebest of both worlds, a fast handheld device full of accessories
and communication options, and a simple durable interface
to sensors on your machines.

Flexible handheld options
Choose from multiple handheld devices
to meet your form factor, screen size, andhazardous rating needs. Update yourhandheld at any time, load the System 1Collector app onto the new device and getcollecting.

So easy to carry

Carry the data acquisition module using ashoulder strap or a belt clip. Sensors mount onto sensor
keepers. Hold the industrial smartphone or tablet in yourhand, on a belt clip, or on a neck strap. So you can focus on
your job and move safely between machines.

Powerful and connectedVibration recordings are viewed and stored on the handheld
device. It runs the Android OS, so you can use it like any
smartphone or tablet to take pictures and video, send
text messages, check email, make phone calls, track yourlocation, create SOS alerts, and run other enterprise apps. Try
doing all that on an old school data collector!

High performancedata acquisition

Unlike most products on the market,
the intrinsically safe SCOUT devices
retain the same high performanceand long battery life as their non-rated COMMTEST brethren.
Ten hours of continuous use, while supporting up to: 80
kHz Fmax, 12,800 line spectral resolution, 6-Pack super fastmeasurements, 4 channel simultaneous sampling, optionaltachometer. And the firmware updates automatically fromthe handheld. Amazing!

System 1 software,your plantwide solution

Use System 1* to prepare routes, analyze
data, detect long-term trends, set anddetect alarms. Then synchronize datawith your SCOUT handheld using WiFi, filetransfer, cellular network, or a USB cable. But
System 1 doesn’t stop there, it’s the same powerful package
that’s coupled with a broad range of Bently Nevada condition
monitoring and protection products. One source of the truth,
modern and plantwide.

Specification summary (For full details see datasheets: SCOUT 109M0226, COMMTEST 109M0792)
Sensor input simultaneous samplingSensors

Compatible sensor types
Base accuracy

Frequency range (±3dB)

1 to 128 averages, 0 to 87.5% overlap

400 to 3200 or 12800 depending on model & chans
Classic algorithm: 23 band options
Peak algorithm: 4 band options

Spec and Wfm for each of: Low Freq, High Freq, DemodBluetooth

BT 4.0 Classic (for communication with handheld)

Battery type

7.4V, 4000mAh, li-Ion, factory replaceable

Size5.1” x 5.1” x 1.4” (130 x 130 x 35 mm)

Operating time

WeightRuggedness drop and sealing
Environmental
Hazardous area rating
Data collector warranty> 10 hours (typical on-route data collection)

SCOUT: 1.5 lb, (680 g) COMMTEST: 1.3 lb (590 g)
4 °F to 140 °F (-20 to 60 °C)
122 °F (50°C) max in Ex haz areas

MIL-STD-810F 4' (1.2 m) drops onto concrete, IP65
SCOUT220-IS and SCOUT240-IS:
IECEx & ATEX, Zone 1 & 21, Ex ia [ia] IIC T4 Ga
CSA Class I & II, Div ABCD & FG
COMMTEST220 & COMMTEST240: non-rated
5 years

Numerous options

Smartphones and tablets from ecom and isafe

Wireless and GPS

4G/LTE, WiFi, BT 4.2, NFC, GPS ..

Screen size range
Cameras
CPUBatteryWeight

MemoryRuggedness

Hazardous area rating
Handheld warranty

Copyright 2021 Baker Hughes Company. All rights reserved.
(04/2021)

3 rpm to 1,200,00 rpm (0.05 Hz to 20 kHz)

AveragingOperating temp

BHCS34047		

Manual or Automatic

25 Hz to 80 kHz (28 options), orders based 1X to 999X

6Pack

Handheld options

Laser, contact, TTL, Keyphasor

Fmax options

Demodulation

Mechanical

1 Hz to 70 kHz (0 Hz to 70 kHz when DC coupled)

5V, 40mA. Rated Zone 2, 22 and Div 2

Spectral resolution

Battery

AC signals: ± 1% of reading, DC level: ± 1% of full scale

Power output
Speed range

Communication

Accelerometer, velocity, displacement, voltage.

3 alternatives, from non-rated to IS rated

Threshold and hysteresis

SpectraSCOUT240-IS & COMMTEST240: 4 channels

Optional laser tachs

Compatible tach types

Tachometer

SCOUT220-IS & COMMTEST220: 2 channels

5” to 8”5 to 12 MP

1.2 to 2 GHz, 4 to 8 cores

3600mAh to 8400 mAh, Li-Ion
325 g to 1250 g

Up to 3 GB RAM, 32 GB ROM, 256 GB card
Up to IP68, -20 to 60 C (-4 to 140°F)

From non-rated to Zone 1/21, Class 1 Div 1
3 years

bently.com