Heavy Duty Pressure Transducers

MLH Series, 6 bar to 550 bar | 50 psi to 8000 psi

32328895

Issue B

Datasheet
FEATURES• Pressure range of 6 bar to 550 bar | 50 psi to 8,000 psi• ±0.25% full scale accuracy Best Fit Straight Line (BFSL)• Total Error Band (TEB) as low as 2 %FSS
• Compensated and operating temperature range of -40°C to
125°C [-40°F to 257°F] (see Table 4)• Less than 2 ms response time
• Fully temperature compensated, calibrated and amplified• Rated IP65 or better for protection from most harshenvironments• RoHS, CE, UL Component Recognition for USA and Canada:
File No. E258956
DESCRIPTION
The MLH Series Heavy Duty Pressure Transducers combine
Application Specific Integrated Circuit (ASIC) technology
with a media-isolated, metal diaphragm design. This digitallycompensated transducer offers value and performance,
making it a suitable pressure sensing solution for demanding
applications. Industry standard electrical connectors andpressure ports are offered for enhanced reliability and user
flexibility.
The MLH Series has six standard output options:• Ratiometric: 0.5 Vdc to 4.5 Vdc from 5 Vdc excitation• Current: 4 mA to 20 mA• Regulated: 1 Vdc to 6 Vdc• Regulated: 0.25 Vdc to 10.25 Vdc• Regulated: 0.5 Vdc to 4.5 Vdc• Regulated: 1 Vdc to 5 Vdc

VALUE TO CUSTOMERS• Total Error Band (TEB) as low as 2 %FSS from -40°C
to 125°C: Provides indication of the transducer’s truemeasurement performance over a specified temperature
range; small error promotes system uptime and efficiency. (SeeFigure 1.)• All metal wetted parts allow for potential use in a wide variety
of fluid applications.
• No internal elastomeric seals eliminate O-ring compatibility
issues.
• Amplified outputs reduce the need to purchase external
amplifiers.
• Input reverse voltage protection guards against mis-wiring.
• Less than 2 ms response time provides more accurate, highspeed measurement.

Sensing and Internet of Things

POTENTIAL APPLICATIONS• Industrial: Compressors, cylinder tank pressure, HVAC,
hydraulics, oil and gas, refrigeration, water distribution• Transportation: Multiple applications including braking andalternative fuels• Medical: Anesthesia delivery machines, blood analyzers,
chemistry analyzers, gas chromatography, oxygen
concentrators, and ventilators
PORTFOLIO
The MLH Series joins the PX3 Series, PX2 Series, SPT Series
heavy duty pressure transducers and the 13 mm Series and
19 mm Series heavy duty pressure transducers.
Figure 1. Total Error Band (TEB) Definition

Sources of Error
OffsetFull Scale Span
Pressure Non-Linearity
Pressure Hysteresis
Pressure Non-Repeatability
Thermal Effect on Offset
Thermal Effect on Span
Thermal Hysteresis

Accuracy
BFSL

Total
Error
Band

Heavy Duty Pressure Transducers

MLH Series, 6 bar to 550 bar | 50 psi to 8000 psi
Table 1. Pressure Range Specifications1 (At 25°C [77°F] and at rated excitation unless otherwise specified.)bar
psi
OperatingPressure

Proof
Pressure

Burst
Pressure

OperatingPressure

Proof
Pressure

Burst
Pressure2

6

18

60

50

150

500

10

30

100

100

300

1000

16

48

160

150

450

1500

25

75

250

200

600

2000

40

80

400

250

750

2500

60

120

600

300

900

3000

100

200

1000

500

1500

5000

160

320

1600

1000

2000

10000

250

500

2068

2000

4000

20000

350

700

2068

3000

6000

30000

500

750

2068

5000

7500

30000

550

825

2068

8000

12000

30000

Comparable metric units follow same proof and burst specifications.
2
Bonded washer seal used on G 1/8 (DIN 3852-2) port is limited to 25000 psi burst.
1

Table 2. Electrical Specifications (At 25°C [77°F] unless otherwise noted.)
Output Signal
Characteristic

Zero output

RatiometricA

Current
B

Regulated
C

Regulated
D

Regulated
E

RegulatedG

0.5 Vdc

4 mA

1 Vdc

0.25 Vdc

0.5 Vdc

1 Vdc

5 Vdc
(1 Vdc to 6 Vdc)

10 Vdc(0.25 Vdc to 10.25 Vdc)

4 Vdc
(1 Vdc to 5 Vdc)

Excitation

5 Vdc(6 Vdc max.)1

9.5 Vdc to30 Vdc2

8 Vdc to30 Vdc2

14 Vdc to
30 Vdc2

4 Vdc(0.5 Vdc to 4.5
Vdc)
7 Vdc to30 Vdc2

Supplycurrent

4 mA typ.,
8 mA max.

N/A

5 mA typ.,
17 mA max.

5 mA typ.,
17 mA max.

5 mA typ.,
17 mA max.

5 mA typ.,
17 mA max.

Source(nominal)

1 mA

N/A

1 mA

1 mA

1 mA

1 mA

Sink
(nominal

1 mA
at zero output

N/A

A1 mA
at zero output

1 mA
at zero output

1 mA
at zero output

1 mA
at zero output

Supplyrejection ratio

90 dB

90 dB

90 dB

90 dB

90 dB

90 dB

Outputimpedance

25 Ohm max.

N/A

25 Ohm max.

25 Ohm max.

25 Ohm max.

25 Ohm max.

Full scalespan (FSS)

1
2

4 Vdc
16 mA
(0.5 Vdc to 4.5 Vdc) (4 mA to 20 mA)

8 Vdc to30 Vdc2

Maintains ratiometricity at 5 ±0.25 Vdc excitation. Product can tolerate 6 Vdc excitation without damage.
See Figures 2 and 3 for more information regarding maximum excitation voltage vs. operating temperature.

Table 3. Pressure Reference Types
Pressure Reference
Sealed gage1

Description
Output is proportional to the difference between applied pressure and a built-in fixed reference to
1 atmA, where the minimum operating pressure is set to 14.7 psiA (1 atmA).

Transducer measures pressure relative to ambient pressure. Output is proportional to the difference
Vented gage (relative) 2 between applied pressure and atmospheric (ambient) pressure, where the minimum operatingpressure is set to atmospheric pressure.
2

Sensing and Internet of Things

Heavy Duty Pressure Transducers

MLH Series, 6 bar to 550 bar | 50 psi to 8000 psi
Table 4. Environmental and Mechanical Specifications (At 25°C [77°F] unless otherwise noted.)
Characteristic

Parameter

Material in contact with media:
portdiaphragm

stainless steel 304L
Haynes 214 alloy

Housing material

black plastic – Amodel AS-4133 HS – PPA

Weight (typical for Metri-Pack 150 and 1/8 NPTpressure port types)

57.0 g [2.0 oz]

Shock

100 g peak [11 ms]

Vibration

MIL-STD-810C, Figure 514.2-5, Curve AK, Table 514.2-V,
Random Vibration Test (overall g rms = 20.7 min.)Compensated and operating temperature range:
0.5 Vdc to 4.5 Vdc ratiometric output
all regulated and 4 mA to 20 mA outputs-40°C to 125°C [-40°F to 257°F]-40°C to 125°C [-40°F to 257°F] (See Figures 2 and 3 for operating area details.)

Storage temperature range-40°C to 125°C [-40°F to 257°F]

Approvals

RoHS, CE, UL Component Recognition for USA and Canada: File No. E258956

Table 5. Performance Specifications (At 25°C [77°F] unless otherwise noted.)
Characteristic

Parameter

Response time

<2 ms

Accuracy :>100 psi<100 psi±0.25 %FSS
±0.50 %FSS

1

Total Error Band2:
Vented gage (relative):<300 psig>300 psig
Sealed gage:
without L, M, P electrical connector types:
100 psis to 299 psis (-40°C to 85°C [-40°F to 185°F])
100 psis to 299 psis (>85°C to 125°C [>185°F to 257°F])>300 psis (-40°C to 125°C [-40°F to 257°F])
with L, M, P electrical connector types:
100 psis to 299 psis (-40°C to 65°C [-40°F to 149°F])
100 psis to 299 psis (>65°C to 125°C [>149°F to 257°F])>300 psis (-40°C to 65°C [-40°F to 149°F])>300 psis (>65°C to 125°C [>149°F to 257°F])±3 %FSS±2 %FSS±3 %FSS±10 %FSS±2 %FSS±10 %FSS±15 %FSS±5 %FSS±15 %FSS

Includes pressure non-linearity (BFSL), pressure hysteresis and pressure non-repeatability. Thermal errors are not included. (SeeFigure 1.)
2
Includes offset error, full scale span error, pressure non-linearity (BFSL), pressure hysteresis, pressure non-repeatability, thermal
effect on offset, thermal effect on span, and thermal hysteresis. (See Figure 1.)
1

Sensing and Internet of Things

3

Heavy Duty Pressure Transducers

MLH Series, 6 bar to 550 bar | 50 psi to 8000 psi
Operating Temperature (°C)

Figure 2. Regulated Output Operating and Temperature Compensation
125
124
123
122
121
120
119
118
117
116

Operating and temperaturecompensated area
for regulated output

6

8

10

12

14

16

18

20

22

24

26

28

30

Supply Voltage (Vdc)
Note: The dot indicates the maximum operating temperature of 125°C [257°F] with a 24 V supply.

Operating Temperature (°C)

Figure 3. Current Output Operating and Temperature Compensation
125
120
115
110
105
100
95
90
85
80
75
70
65
60

10 Ohmload

250 Ohmload

Operating and temperaturecompensated area
for current output
6

8

10

12

14

16

18

20

22

24

26

28

30

Supply Voltage (Vdc)
Note: The operating area is extended with a 250 Ohm resistor. Higher loads extend the operating area. The dot indicates the
maximum operating temperature when using a 24 V supply and a 250 Ohm resistor.

CAUTIONPRODUCT DAMAGE DUE TO MECHANICAL ISSUES• Ensure torque specifications are determined for the specific application. Values provided are for reference only. (Mating
materials and thread sealants can result in significantly different torque values from one application to the next.)• When using mating parts made of stainless steel, use a thread sealant with anti-seize properties to prevent thread galling.
Ensure the sealant is rated for the application.
• Use appropriate tools (such as an open ended wrench or deep well socket) to install transducers.
• Always hand-start transducers into the hole to prevent cross threading and damage.
• Ensure that torque is not applied to the electrical connector.
• Ensure that the proper mating electrical connector with a seal is used to connect the transducer. Improper or damaged
seals can compromise ingress protection, leading to short circuits.
Failure to comply with these instructions may result in product damage.

CAUTIONPRODUCT DAMAGE DUE TO PARTICULATES• Ensure that a filter is used upstream of the transducer to keep media flow free of larger particulates and increased humidity.
All MLH Series transducers are dead-ended devices; particulate accumulation and condensing moisture may affecttransducer output.
• It is recommend that the transducer be positioned with the port facing downwards; any particulates in the system are lesslikely to enter and settle within the pressure transducer if it is in this position.
• Ensure that the media does not create a residue when dried. Build-up inside the transducer may affect transducer output;
rinsing of a dead-ended transducer is potentially difficult and has limited effectiveness in removing residue.
Failure to comply with these instructions may result in product damage.

4

Sensing and Internet of Things

Heavy Duty Pressure Transducers

MLH Series, 6 bar to 550 bar | 50 psi to 8000 psi
Figure 4. Nomenclature and Order GuideE
For example, MLH150PSB01B defines an MLH Series Heavy Duty Pressure Transducer, 150 psi pressure range, sealed
gage pressure reference, Metri-Pack 150 electrical connector type, 1/4-18 NPT pressure port type, 4 mA to 20 mA output.

MLH

150

P

S

B

01

B
Output Transfer Function

SeriesHeavy
Duty

A Ratiometric: 0.5 Vdc to 4.5 Vdc

Transducer1

B Current: 4 mA to 20 mA
C Regulated: 1 Vdc to 6 Vdc
D Regulated: 0.25 Vdc to 10.25 Vdc
E Regulated: 0.5 Vdc to 4.5 Vdc
G Regulated: 1 Vdc to 5 Vdc

from 5 Vdc excitation

MLH Pressure
Pressure Range
bar

006
010
016
025
040
060
100
160
250
350
500
550

6
10
16
25
40
60
100
160
250
350
500
550

psi

050 50100 100
150 150200 200
250 250300 300
500 500
01K 1000
02K 2000
03K 3000
05K 5000
08K 8000

Electrical Connector Type
Metri-Pack 1503

B

Cable harness,
1 meter cable
length3

L

Hirschmann

Cable harness,
3 meter cable
length33

C

M

Unit
B bar
P psi
Pressure Reference
G Vented gage (relative)
S

M12x1(Brad Harrison micro)

Flying leads

D

01

06

11

13

19

DIN 43650-C, 8 mm maleG54

1/8-27 NPT(ANSI B1.20.1)

55

7/16-20 UNF
1/4 in 45º
09 Flare Female
Schrader withdepressor

P

Sealed gage2

Metri-Pack 150 mating connector3

Pressure Port Type
1/4-18 NPT(ANSI B1.20.1)

50
Deutsch DTM04-3P(integral)

51

56

1/2-14 NPT(ANSI B1.20.1)

57

R 1/4-19 BSPT(ISO 7-1)

58

R 1/8-28 BSPT(ISO 7-1)

59

3/8-24 UNF with O-ring seal
(SAE J1926-2)5

7/16-20 UNF with O-ring seal
(SAE J1926-2)5

M10x1 with O-ring seal(ISO 6149-2)5

M12x1.5 with O-ring seal(ISO 6149-2)5

M14x1.5 with O-ring seal(ISO 6149-2)5

M16x1.5 with O-ring seal(ISO 6149-2)5

M18x1.5 with O-ring seal(ISO 6149-2)5

M20x1.5 with O-ring seal(ISO 6149-2)5

G1/8-28 BSPP
with bonded washer

60 (DIN 3852-2)5, 6

61

G1/4-19 BSPP
with bonded washer(DIN 3852-2)5, 6

62

G1/8-28 BSPP
with elastomeric seal(ISO 1179-2)5

AMP Superseal 1.54

T

52

1/2-20 UNF with O-ring seal
(SAE J1926-2)5H
53

9/16-18 UNF with O-ring seal
(SAE J1926-2)5

G1/4-19 BSPP
with elastomeric seal

63 (ISO 1179-2)5

Not all catalog listing combinations are available. Custom products are available. Please contact Honeywell.
Sealed gage option is only available in pressure ranges at or above 8 bar | 100 psi.
Metri-Pack 150 mating connectors with shielded cable and 22 AWG wire are available from Honeywell. Order part no. 3685301 for 1 m [3.2 ft] cable length and part no. 3685302 for 3 m[9.8 ft] cable length.
4
Available with Output Transfer Function A only.
5
Seal accessory is included with the sensor. Seal accessory materials and specifications are: O-ring (nitrile, durometer 90, temperature range -30°C to 125°C), bonded washer (steel outerring with nitrile inner ring, temperature range -30°C to 125°C), elastomeric seal (nitrile, temperature range -30°C to 125°C).
6When using the bonded washer accessory supplied by Honeywell, the customer’s female port connection design should follow DIN 3852-1: 2000-11 size large. If compliance to
DIN 3852-1:2014-05 is required, contact Honeywell for alternate options.
1
2
3

Sensing and Internet of Things

5

Heavy Duty Pressure Transducers

MLH Series, 6 bar to 550 bar | 50 psi to 8000 psi
Figure 5. Electrical Connector Type Mounting Dimensions (For reference only. mm/[in].)
General product marking“H” + “P” = total nominal product length

Catalog listing

27,0 Hex flat to flat
[1.06]

“P”

“H”

B

C

Metri-Pack 150

D

Hirschmann

Connector: Delphi 12078088
Mating connector: Delphi
12110192
IP rating1: IP65 (all versions)

Date code

M12x1 (Brad Harrison micro)

Connector: DS/EN 61076-2Connector: Hirschmann G
101 (IEC 61076-2-101)
series, 4-pole (small cubic
Mating connector: 4 POS
connector)
Mating connector: Hirschman Type D
IP rating1: IP65, IP67 (sealed
G4WF or equivalent
IP rating1: IP65 (all versions) gage versions), IP65 (vented
gage versions)G

DIN 43650-C, 8 mm maleConnector: DIN 43650C, 8 mm
Mating connector: DIN
43650C, 8 mm
IP rating1: IP65 (all versions)

Pin

Voltage
Output(3-wire)Current
Output(2-wire)Pin

Voltage
Output(3-wire)Current
Output(2-wire)Pin

Voltage
Output(3-wire)Current
Output(2-wire)Pin

Voltage
Output(3-wire)Current
Output(2-wire)A+excitation+excitation

1

noconnection

noconnection

1+excitation+excitation1+excitation+excitationB

commonnoconnection

2

outputnoconnection

2common-excitation(return)2

common-excitation(return)C

output-excitation(return)3

common-excitation(return)3

outputnoconnection

3

4+excitation+excitation

4

noconnection

noconnection

4

no
no
connenction connection
output

Pin C
Pin B

Pin A

Pin 3
Pin 1

Pin 4

Pin 4

Pin 3

Pin 2

No connection

Pin 2

Pin 2

Pin 1

Pin 3

Pin 1
Pin 4

ø16,5117,0[0.67][0.46]

ø16,0

ø10,0

[0.63]“H” = 41,7
[1.64]

6

Hole(Ventedgage
versions
only)

Sensing and Internet of Thingsnoconnection“H” = 32,6
[1.28][0.39]

Hole(Ventedgage
versions
only)“H” = 33,5
[1.32]

Hole(Ventedgage
versions
only)“H” = 30,1
[1.19]

Hole(Ventedgage
versions
only)Heavy Duty Pressure Transducers

MLH Series, 6 bar to 550 bar | 50 psi to 8000 psi
Figure 5. Electrical Connector Type Mounting Dimensions (continued)

L

H

AMP Superseal 1.5

TP

Cable Harness, 1 Meter

M

Flying Leads

Deutsch DTM04-3P(integral)

Connector: 24 AWG with TPE
jacketMating connector: Flying
leads
IP rating1: IP65, IP67, IP69K(sealed gage versions), IP65
(vented gage versions)

Connector: 6 in flying leads
with 20 AWG wire and PE
insulation
IP rating1: IP65, IP67, IP69K(sealed gage versions), IP65
(vented gage versions)

Connector: DTM04-3P
Mating connector: DTM063S with wedgelock WM-3S
IP rating1: IP65, IP67, IP69K(sealed gage versions), IP65
(vented gage versions)

Cable Harness, 3 Meter
Connector: Amp 282087
Mating connector: Amp
282087IP rating1: IP65 (all versions)

Pin

Voltage
Output(3-wire)

Wire
Color

Voltage
Output(3-wire)Current
Output(2-wire)

Wire
Color

Voltage
Output(3-wire)Current
Output(2-wire)Pin

Voltage
Output(3-wire)1

commonred+excitation +excitation

red+excitation +excitation1+excitation +excitation

2

outputwhite

common-excitation(return)2

common-excitation(return)

outputnoconnection(black wire
removed)3

outputnoconnection

3+excitation

black

outputcommon-excitation
white(return)noconnection(black wire black
removed)Current
Output(2-wire)

White

Pin 3

Black
(Used for voltage
ouptut only)

White

Red

Pin 1

Pin 2

Pin 2

Black
(Used for voltage
ouptut only)

Pin 1

13,8
[0.54]“H” = 41,0[1.61]

1
2

13,4[0.53][0.43][0.43]“H” = 35,3
[1.39]

Pin 3

ø10,8

ø10,8

Hole
(Ventedgage
versions
only)

Red

Hole
(Ventedgage
versions
only)“H” = 35,3
[1.39]“H” = 42,6
[1.68]
Hole(Ventedgage
versions
only)

Hole
(Ventedgage
versions
only)

IP rating is determined by the electrical connector type chosen.
Three-wire cable is required for ratiometric and regulated voltage outputs; two wire cable is required for current output.

Sensing and Internet of Things

7

Heavy Duty Pressure Transducers

MLH Series, 6 bar to 550 bar | 50 psi to 8000 psi
Figure 6. Pressure Port Type Mounting Dimensions (For reference only: mm/[in].)1
General product marking“H” + “P” = total nominal product length

Catalog listing

27,0 Hex flat to flat
[1.06]

Date code

“P”

“H”

01

09

06

7/16-20 UNF 1/4 in 45º
Flare Female Schrader with depressor

1/4-18 NPT

1/8-27 NPT

Seal: Pipe thread
Mating geometry: ANSI B1.20.1-2013
Installation torque: 2 to 3 turns fromfinger tight

Seal: Pipe thread
Mating geometry: ANSI B1.20.1-2013
Installation torque: 2 to 3 turns fromfinger tight“P” = 20,1
[0.79]

15,2[0.60]

11

20,1[0.79]

50

3/8-24 UNF (SAE J1926-2)
Seal1: O-ring
Mating geometry: SAE J1926-1
Installation torque: 10 N m [7.4 ft-lb]

13,5[0.53]

53

12,7[0.50]

19

Seal: Pipe thread
Mating geometry: ISO 7-1
Installation Torque: 2 to 3 turns fromfinger tight“P” = 19,6
[0.77]“P” = 17,5[0.69]

R1/8-28 (BSPT)

R1/4-19 (BSPT)

Seal: Pipe thread
Mating geometry: ANSI B1.20.1-2013
Installation Torque: 2 to 3 turns fromfinger tight“P” = 18,3
[0.72]

10,2[0.40]

13

1/2-14 NPT“P” = 24,9[0.98]“P” = 15,0[0.59]

Seal: 45º cone
Mating geometry: SAE J512
Installation torque: 20 Nm [14.8 ft-lb]

14,7[0.58]

51

Seal: Pipe thread
Mating geometry: ISO 7-1
Installation Torque: 2 to 3 turns fromfinger tight“P” = 15,0[0.59]

10,2[0.40]

52

7/16-20 UNF (SAE J1926-2)

1/2-20 UNF (SAE J1926-2)

Seal1: O-ring
Mating geometry: SAE J1926-1
Installation torque: 20 N m [14.8 ft-lb]

Seal1: O-ring
Mating geometry: SAE J1926-1
Installation torque: 40 N m [29.5 ft-lb]“P” = 19,8
[0.78]

15,0[0.59]

54“P” = 19,8
[0.78]

15,0[0.59]

55

9/16-18 UNF (SAE J1926-2)

M10x1.0 (ISO 6149-2)

M12x1.5 (ISO 6149-2)

Seal1: O-ring
Mating geometry: SAE J1926-1
Installation torque: 45 N m [33.2 ft-lb]

Seal1: O-ring
Mating geometry: ISO 6149-2
Installation torque: 20 N m [14.8 ft-lb]

Seal1: O-ring
Mating geometry: ISO 6149-2
Installation torque: 45 N m [33.2 ft-lb]“P” = 20,8[0.82]

8

16,0[0.63]

Sensing and Internet of Things“P” = 15,9[0.63]

11,1[0.44]“P” = 18,3
[0.72]

13,5[0.53]

Heavy Duty Pressure Transducers

MLH Series, 6 bar to 550 bar | 50 psi to 8000 psi
Figure 6. Pressure Port Type Mounting Dimensions (continued)

56

57

58

M14x1.5 (ISO 6149-2)

M16x1.5 (ISO 6149-2)

M18x1.5 (ISO 6149-2)

Seal1: O-ring
Mating geometry: ISO 6149-1
Installation torque: 45 N m [33.2 ft-lb]

Seal1: O-ring
Mating geometry: ISO 6149-1
Installation torque: 55 N m [40.6 ft-lb]

Seal1: O-ring
Mating geometry: ISO 6149-1
Installation torque: 70 N m [51.6 ft-lb]“P” = 18,3
[0.72]

13,5[0.53]

59“P” = 19,8
[0.78]“P” = 21,3
[0.84]

15,0[0.59]

60

16,5[0.65]

61

M20x1.5 (ISO 6149-2)

G1/8-28 (BSPP) DIN 3852-2

G1/4-19 (BSPP) DIN 3852-2

Seal1: O-ring
Mating geometry: ISO 6149-1
Installation torque: 80 N m [59.0 ft-lb]

Seal1: Bonded washer
Mating geometry with bonded washer:
DIN3852-1; 2000-11, large
Installation torque: 20 N m [14.8 ft-lb]

Seal1: Bonded washer
Mating geometry with bonded washer:
DIN 3852-1:2000-11, large
Installation torque: 50 N m [36.9 ft-lb]“P” = 19,0[0.75]“P” = 14,6
[0.57]“P” = 21,3
[0.84]

16,5[0.65]

2,0[0.08]

Bondedwasher

[0.41]

62“P” =14,3[0.56]

Elastomeric
seal

ø8,4

[0.33]

120°

ø11,9

[0.47]

1,0[0.04]

1

Bondedwasher

[0.63]

ø13,7[0.54]

Bonded washer

ø20,6[0.81]

63

G1/4-19 (BSPP) ISO 1179-2
Seal1: ElastomericMating geometry: ISO 1179-1
Installation torque: 50 N m [36.9 ft-lb]“P” = 18,8[0.74]

9,5[0.37]

0,5[0.02]

2,0[0.08]

Bonded washer

ø10,4 ø15,9

G1/8-28 (BSPP) ISO 1179-2
Seal1: ElastomericMating geometry: ISO 1179-1
Installation torque: 20 N m [14.8 ft-lb]

14,1[0.56]

9,7[0.38]

14,0[0.55]

Proper assembly ofseal to stud end

0,8[0.03]

Proper assembly ofseal to stud end

Elastomeric
seal

ø11,6

[0.46]

120°

ø16,5

[0.65]

1,5[0.06]

Seal accessory included with transducer. Seal accessory material and specifications: O-ring (nitrile, durometer 90, temperature range -30 ° C
to 125° C), bonded washer (steel outer ring with nitrile inner ring, temperature range -30 ° C to 125° C), elastomeric seal (nitrile, temperature
range -30 ° C to 125° C).

Sensing and Internet of Things

9ADDITIONAL INFORMATION

The following associated literature is available on the
Honeywell web site at sensing.honeywell.com:

WARNINGPERSONAL INJURY
DO NOT USE these products as safety or emergency stop
devices or in any other application where failure of the
product could result in personal injury.

• Product line guide• Product range guide• Product installation instructions

Failure to comply with these instructions could result indeath or serious injury.

• Product CAD models

WARNINGMISUSE OF DOCUMENTATION••

The information presented in this datasheet is forreference only. Do not use this document as a productinstallation guide.
Complete installation, operation, and maintenanceinformation is provided in the instructions supplied witheach product.

Failure to comply with these instructions could result indeath or serious injury.

Warranty/Remedy
Honeywell warrants goods of its manufacture as being freeof defective materials and faulty workmanship during theapplicable warranty period. Honeywell’s standard productwarranty applies unless agreed to otherwise by Honeywell in
writing; please refer to your order acknowledgement or consult
your local sales office for specific warranty details. If warranted
goods are returned to Honeywell during the period ofcoverage, Honeywell will repair or replace, at its option, withoutcharge those items that Honeywell, in its sole discretion, finds
defective. The foregoing is buyer’s sole remedy and is in lieu
of all other warranties, expressed or implied, including thoseof merchantability and fitness for a particular purpose. In no
event shall Honeywell be liable for consequential, special, or
indirect damages.
While Honeywell may provide application assistancepersonally, through our literature and the Honeywell web site, it
is buyer’s sole responsibility to determine the suitability of the
product in the application.

For more information
Honeywell Sensing and Internet ofThings services its customers through aworldwide network of sales offices anddistributors. For application assistance,
current specifications, pricing or thenearest Authorized Distributor, visit
sensing.honeywell.com or call:
Asia Pacific+65 6355-2828
Europe 		+44 (0) 1698 481481
USA/Canada+1-800-537-6945

Specifications may change without notice. The information we
supply is believed to be accurate and reliable as of this writing.
However, Honeywell assumes no responsibility for its use.

Honeywell Sensing and Internet of Things
9680 Old Bailes Road
Fort Mill, SC 29707
www.honeywell.com

32328895-B-EN | B | 03/18© 2018 Honeywell International Inc. All rights reserved.