MIP SERIES

32346591

Issue B

Heavy Duty, Media-Isolated Pressure Transducers
1 bar to 60 bar | 15 psi to 870 psi

DESCRIPTIONDIFFERENTIATIONThe MIP Series offers a heavy duty,
media-isolated pressure transducer in acompact, stainless steel construction for
use with a wide range of media including
aggressive fluids and water. The MIP
Series provides a cost-competitive
solution for wide-ranging potential
applications in tough environments.

• Diagnostics: Beneficial in applications
where the sensor functionality andthe need to know internal or external
failure modes is critical.
• Great customer value: Multiple
configuration possibilities provide
flexibility of use in the application withno up front NRE or tooling charges.
• Durable: Provides the toughenvironmental capabilities needed,
including long-term stability, insulation
resistance and dielectric strength,
external freeze-thaw resistance and
EMC performance.

VALUE TO CUSTOMERS• Total Error Band (TEB) ±0.75 %FSS
(Full Scale Span) from -40°C to125°C: Provides true measurement
performance over the compensatedtemperature range; small error
promotes system uptime andefficiency (see Figure 4).
• EMC performance: Operates reliably
in the presence of electromagnetic
fields, such as wireless signals, RFcommunication, and electrical devices.
• Hermetically welded design supports
almost any media without the use of aninternal seal. The sensors are designed tobe used in harsh environments that see
aggressive media.

FEATURES

Certified toNSF/ANSI/CAN 61

TABLE 1. POTENTIAL APPLICATIONSINDUSTRY
Industrial:
pumps
compressors
process

MEDIA
water, hydraulic fluids
compressed air
food, beverage, oil, gas, steam• Rugged, stainless steel construction• Ratiometric output: 0.5 Vdc to 4.5 Vdc• Operating temperature: -40°C to 125°C• Total Error Band up to ±0.75 %FSS(-40°C to 125°C)• Industry-leading accuracy: ±0.15%FSS BFSL
• Long term stability: ±0.25 %FSS
• Radiated immunity: 100 V/m• High burst pressure: 207 bar [3000 psi]
• Drinking water approval: NSF/ANSI/CAN 61• CE, RoHS, REACH compliant• Mis-wiring protection• Over voltage protection ±40 Vdc

PORTFOLIO
Honeywell offers a variety

HVAC/Rrefrigerants (butane, propane, ammonia, CO2, R134A, R407C,
R410A, R448A/Solstice® N40, R32 and R1234ze, R1234yf,
glycol + watertransducers for potential use inTransportation

gasoline, diesel fuel, engine oil, brake fluid, coolants, CNG

industrial and transportation

MedicalO2, N2, CO2, N2O, airapplications. To view the entireof heavy duty pressure

product portfolio, click here.

HEAVY DUTY, MEDIA-ISOLATED PRESSURE TRANSDUCERS, MIP SERIESTABLE 2. ELECTRICAL SPECIFICATIONS (AT 25°C [77°F] AND UNDER UNLESS OTHERWISE NOTED.)CHARACTERISTIC

PARAMETER

Supply voltage

5.0 ±0.25 Vdc

Nominal output transfer function (5 Vdc supply)

0.5 Vdc to 4.5 Vdc

Over/reverse voltage±40 Vdc

Current consumption

6.5 ±1 mA

Short circuit protection

yes

TABLE 3. PERFORMANCE SPECIFICATIONS (AT 25°C [77°F] AND UNDER UNLESS OTHERWISE NOTED.)CHARACTERISTIC

PARAMETER

Operating temperature range-40°C to 125°C [-40°F to 257°F]

Total Error Band(-40°C to 125°C [-40°F to 257°F])±1.0 %FSS (<10 bar)2±0.75 %FSS (>10 bar)

Accuracy BFSL3±0.15 %FSS

Long term stability (1000 hr, 25°C [77°F])±0.25 %FSS

Minimum output resolution

0.03 %FSS

Response time

1 ms (10% to 90% step change in pressure)Startup time

<7msEMC rating (CE Conformity):
surge immunity (all leads)
electrostatic discharge
radiated immunity
fast transient burst
immunity to conducted disturbances
radiated emissions±1 kV line to ground per IEC 61000-4-5±4 kV contact, ±8 kV air per IEC 61000-4-2
10 V/m (80 MHz to 1000 MHz) per IEC 61000-4-3±1 kV per IEC 61000-4-4
3 V (150 kHz to 80 MHz) per IEC 61000-4-6
40 dBμV (30 MHz to 230 MHz), 47 dBμV (230 MHz to 1000 MHz) per CISPR 11

Radiated immunity

100 V/m (200 MHz to 2 GHz) per ISO 11452-2

Insulation resistance>100 MΩ at 1 kVdc (60 s)

Dielectric strength<1 mA at 500 Vac (60 s)

Load resistance≥2 kΩ (pull up or pull down)

Life>10 million full scale pressure cycles

14Total Error Band: The maximum deviation from the ideal transfer function over the entire compensated temperature and pressure
range. Includes all errors due to offset, full scale span, pressure non-linearity, pressure hysteresis, pressure non-repeatability,
thermal effect on offset, thermal effect on span, and thermal hysteresis (see Figure 4).
2TEB: ±1.5 % FSS above 100°C [212°F] for pressure ratings less than 4 bar [58 psi].
3Accuracy: The maximum deviation in output from a Best Fit Straight Line (BFSL) fitted to the output measured over the pressure
range at 25°C [77°F]. Includes all errors due to pressure non-linearity, pressure hysteresis, and pressure non-repeatability.
4Startup time: The time needed to receive valid output after power up.
1TABLE 4. ENVIRONMENTAL AND MECHANICAL SPECIFICATIONS

2CHARACTERISTIC

PARAMETER

Shock

100 G per MIL-STD-202, Method 213, Cond. C (at 25°C [77°F])

Vibration

20 G sweep, 10 Hz to 2000 Hz (at 25°C [77°F])

Ingress protection

IP67

External freeze/thaw resistance>6 cycles from -30°C to 50°C [-22°F to 122°F]

Wetted materials:
portdiaphragm
external seal for ports

stainless steel 304L
stainless steel 316L
nitrile (-30°C to 100°C [-22°F to 212°F]) (other materials available)

Electrical connector material

PBT 30%GF (UL V-0)

sensing.honeywell.com

HEAVY DUTY, MEDIA-ISOLATED PRESSURE TRANSDUCERS, MIP SERIESFIGURE 1. NOMENCLATURE AND ORDER GUIDEFor example, MIPAN1XX010BSAAX defines an MIP Series Heavy Duty, Media Isolated Pressure Transducer,
Metri-Pack 150, standard (UL V-0) electrical connector type, 1/4-18 NPT pressure port type, 10 bar pressure
range, sealed gage pressure reference, ratiometric: 5 Vs, 10% to 90% Vs output transfer function

MIP A N1 X X 010B S AA X1SpecialSeriesXMIP Media Isolated Pressure

Output Transfer Function

AA

Electrical Connector TypeA

Ratiometric: 5.0 Vs, 10% to 90% Vs

Pressure Reference

Metri-Pack 150,
Standard (UL V-0)

Pressure Range

Pressure Port TypeF1

7/16-20 UNF
1/4 inch 45° Flare
Female Schrader(SAE J512)

G1

G1/4 A-G(ISO 1179-3)

G2

G1/4 A-L(ISO 1179-2)

M1

M12 x 1.5
(ISO 6149-3)

N1

1/4-18 NPT(ANSI/ASME B1.20.1)

N2

1/8-27 NPT(ANSI/ASME B1.20.1)

AS001B 1 bar
002B 2 bar
004B 4 bar
006B 6 bar
008B 8 bar
010B 10 bar
012B 12 bar
016B 16 bar

bar
020B
025B
035B
040B
046B
050B
055B
060B

Absolute
Sealed gage2

psi
20 bar
25 bar
35 bar
40 bar
46 bar
50 bar
55 bar
60 bar

015P
030P
050P
060P
100P
150P
200P
250P
300P

15 psi30 psi50 psi60 psi
100 psi
150 psi
200 psi
250 psi
300 psi

500P 500 psi
600P 600 psi
667P 667 psi
700P 700 psi
750P 750 psi
800P 800 psi
850P 850 psi
870P 870 psi

SpecialX
SpecialX

12

Contact Honeywell Sales for custom configurations.
Sealed gage option only available in pressure ranges at or above 8 bar | 100 psi.

CAUTIONPRODUCT DAMAGE DUE TO MISUSE• Ensure torque specifications are determined for the specific application. Values provided are for reference only. (Matingmaterials and thread sealants can result in significantly different torque values from one application to the next.)• Use appropriate tools (such as an open ended wrench or deep well socket) to install transducers.
• Ensure that the proper mating electrical connector with a seal is used to connect the transducer. Improper or damaged seals
can compromise ingress protection, leading to short circuits.
• Ensure that filters are used upstream of the transducers to keep media flow free of large particulates and condensed moisture.
MIP Series transducers are dead-end devices. Particulate accumulation may clog the port or damage the diaphragm.
• Ensure the transducer is installed with the port facing downwards. Any particulates in the system are less likely to enter andsettle within the transducer if it is in this position.
• Ensure that the media does not create a residue when dried. Build-up of residue inside the transducer may affect transducer
outputFailure to comply with these instructions may result in product damage.

Sensing and Internet of Things

3HEAVY DUTY, MEDIA-ISOLATED PRESSURE TRANSDUCERS, MIP SERIESTABLE 5. PRESSURE RATINGSBAROPERATING
PRESSURE

PSI

OVERPRESSURE1

1 to 36>3 to 1224

BURST
PRESSURE2

OPERATING
PRESSURE

OVERPRESSURE1

15 to 43.5

87207>43.5 to 174

348

BURSTPRESSURE23000>12 to 60
120>174 to 870
1740
Overpressure: The maximum pressure which may safely be applied to the product for it to remain in specification once pressure
is returned to the operating pressure range. Exposure to higher pressures may cause permanent damage to the product.
2
Burst Pressure: The maximum pressure which may be applied without causing escape of pressure media. The product should not
be expected to function after exposure to the burst pressure.

1

FIGURE 2. ANALOG OUTPUT TRANSFER FUNCTION

FIGURE 3. ABSOLUTE VS SEALED GAGE

The transfer function shown here is applicable to a ratiometric
output ranging between 10% Vsupply at null pressure to90% Vsupply at full scale pressure.

Example shown is for 100 psi.
54.5

Compensated Pressure Range
97.5

44.8754.5

90Output (Vdc)

102.50.50.125

Ideal
TotalError
Band

Output (V)Output (% Vsupply)

3.5
32.5
100 psi absolute

2100 psi sealed gage

1.5Pmin.

Output (V) =PressurePmax.

0.8 x Vsupply
x (Pressureapplied - Pmin.) + 0.10 x Vsupply
Pmax.- Pmin.

10.5
0020
14.7

FIGURE 4. TEB COMPONENTS FOR THE MIP SERIES
Total Error Band (TEB) is a single specification that includes
the major sources of sensor error. TEB should not be confusedwith accuracy, which is actually a component of TEB. TEB is themaximum error that the sensor could experience.
Honeywell uses the TEB specification in its datasheet because it is
the most comprehensive measurement of a sensor’s true accuracy.
Honeywell also provides the accuracy specification in order toprovide a common comparison with competitors’ literature that
does not use the TEB specification.
Many competitors do not use TEB—they simply specify theaccuracy of their device. Their accuracy specification, however,may exclude certain parameters. On their datasheet, the errors arelisted individually. When combined, the total error (or what would beTEB) could be significant.
TABLE 6. OUTPUT DIAGNOSTICS CODES

4FAULT CONDITIONANALOG DIAGNOSTIC RAIL

Sensor internal failures

97.5% of Vsupply (See Figure 2.)

Over pressure

97.5% of Vsupply (See Figure 2.)

Under pressure

2.5% of Vsupply (See Figure 2.)

Power or ground loss

high (external pull-up resistor)

Power or ground loss

low (external pull-down resistor)

sensing.honeywell.com

40

60

80

100

Pressure (psi Absolute)

120

140

114.7

Sources of Error
OffsetFull Scale Span
Pressure Non-Linearity
Pressure Hysteresis
Pressure Non-Repeatability
Thermal Effect on Offset
Thermal Effect on Span
Thermal Hysteresis

Accuracy
BFSL

TotalErrorBand[0.87]Pin C

HEAVY DUTY, MEDIA-ISOLATED PRESSURE TRANSDUCERS,MIP SERIES
Pin B(Productmarking)

FIGURE 5. MOUNTING DIMENSIONS (FOR REFERENCE ONLY. MM/[IN]) E

0]

Pinout

Pin A

22,0 A/F[0.87]

Product Marking

Ratiometric Voltage Output
Pin A = Ground
Pin B = V+Pin C = Vout

MIPXXXXXXXXXXXXX

16-Digit product listing

H YYMMDD XXXXX(Productmarking)

Honeywelllogo

5-Digit serial number
6-Digit date

CE

CE Mark

Pin C
Pin B

F1: 7/16-20 UNF 1/4 inch 45° Flare Female Schrader (SAE J512)
Seal: 45° cone
Mating geometry: SAE J512
Installation torque: 17 N m [12 ft-lb]
Weight: 36 g (Product[1.3 oz]marking)

G1: G1/4 A-G (ISO MIPXXXXXXXXXXXXX
1179-3)

Seal: O-ring (included) and retaining ring ISO 1179-3-G1/4
16-Digit product listing(not included)
Mating geometry: ISOH
1179-1YYMMDD XXXXX
Installationtorque: 20 N m [14.7 ft-lb]
5-Digit serialnumber
Honeywell
Ø17,0
Ø17,0
Weight: 33 logo
g [1.1 oz]
Ø17,0[0.67]
Ø17,0 Ø17,0[0.67]

Ø17,0[0.67]

CE[0.67]

35,035,0[1.38][1.38]

Ø17,0[0.67][0.67] [0.67]

CE Mark
35,0[1.38]

35,0[1.38]

MIPXXXXXXXXXXXXX

35,0[1.38]

46,249,046,235,0[1.93][1.82][1.38]

46,2[1.82]5-Digit[1.82] serial number
10,26-Digit
date10,2[0.40] [0.40]

CE[0.40]

CE Mark

6-Digit date

49,0[1.93]
10,210,2[0.40][0.40]

46,2[1.82]

35,010,2[1.38][0.40]

46,2[1.82]
10,2[0.40]

10,2[0.40]

G2: G1/4 A-L (ISO 1179-2)

M1: M12 x 1.5 (ISO 6149-3)
Ø17,0

Ø17,0
Ø17,0 [0.67]
Seal: O-ring (included)[0.67]

Ø17,0
Seal: ISO 9974-2/DIN 3869 profile ring(included)[0.67]
Mating geometry: ISO 1179-1
Installation torque: 20 N m [15 ft-lb]
Weight: 36 g [1.3 oz]
Ø17,0

Ø17,0[0.67]

48,535,0[1.91][1.38]

Ø17,0[0.67]

Ø17,010,2 [0.67][0.40]

Ø17,0[0.67]

49,0[1.93]
35,0
35,010,2[1.38][0.40][1.38]

10,2[0.40]

10,2[0.40]

10,2[0.40]

49,0[1.93]

10,2[0.40]

N1: 1/4-18 NPT

35,0[1.38]
35,046,2[1.38]
35,0[1.82][1.38]

50,0[1.97]
10,2[0.40]

Ø17,0[0.67]

Ø17,0[0.67]

35,0[1.38]
35,0[1.38]
46,2[1.82]
10,2[0.40]

10,2[0.40]
50,0[1.97]

10,2[0.40]

10,2[0.40]

10,2[0.40]

N2: 1/8-27 NPT

Seal: Pipe thread
Mating geometry: ANSI B1.20.1
Installation torque: Two to three turns from finger tight
Weight: 30 g [1.0 oz]

Mating geometry: ANSI B1.20.1
Ø17,0Installation torque: Two to three turns[0.67]
from finger tight
Weight: 38 g [1.3 oz]

48,5[1.91]

50,0
46,2[1.97] 35,0
10,2[1.82][1.38][0.40]

48,5[1.91]
35,0[1.38]

Ø17,0Seal: Pipe thread[0.67]

Ø17,0[0.67]

Ø17,0[0.67]

Ø17,0

Ø17,0[0.67]

35,0[1.38]

35,0[1.38]

49,046,2 [1.93][1.82]

46,2[1.82]

46,2[1.82][1.38]

50,0
48,5 [1.97]
Ø17,0[1.91]
10,2 Ø17,0 [0.67]
10,2 [0.40] [0.67][0.40]

35,0[1.38][0.67]

Mating geometry: ISO 6149-1[0.67]
Installation torque: 20 N m [15 ft-lb]
35,0Weight: 34 g [1.2 oz]

35,0[1.38]
35,0[1.38][0.67]

35,0[1.38]

Ø17,0[0.67]

35,0[1.38]

16-Digit product listing

H YYMMDD XXXXX46,2[1.82]
Honeywelllogo 10,220]

20]

Pin B

Ø17,0[0.67]

Ø17,0[0.67]

Ø17,0[0.67]

Ø17,0[0.67]

Ø17,0[0.67]

35,0[1.38]
35,0[1.38]

10,2[0.40]

10,2[0.40]

50,0[1.97]

46,2[1.82]
10,2[0.40]

35,0[1.38]

46,2[1.82]
10,2[0.40]

10,2[0.40]

Sensing and Internet of Things

5ADDITIONAL MATERIALSWARRANTY/REMEDY

The following associated literature isavailable at sensing.honeywell.com:

Honeywell warrants goods of its
manufacture as being free of defective
materials and faulty workmanshipduring the applicable warranty period.
Honeywell’s standard product warranty
applies unless agreed to otherwise by
Honeywell in writing; please refer toyour order acknowledgment or consult
your local sales office for specificwarranty details. If warranted goods
are returned to Honeywell during theperiod of coverage, Honeywell will
repair or replace, at its option, withoutcharge those items that Honeywell,in its sole discretion, finds defective.
The foregoing is buyer’s sole remedy
and is in lieu of all other warranties,
expressed or implied, including thoseof merchantability and fitness for aparticular purpose. In no event shall
Honeywell be liable for consequential,
special, or indirect damages.

• Product range guide• Application-specific and technicalinformation• CAD Models

While Honeywell may provide
application assistance personally,through our literature and theHoneywell web site, it is buyer’ssole responsibility to determine
the suitability of the product in theapplication.

FOR MORE INFORMATION

Specifications may change without
notice. The information we supply isbelieved to be accurate and reliable
as of this writing. However, Honeywellassumes no responsibility for its use.

Honeywell Sensing and Internet ofThings services its customers through aworldwide network of sales offices anddistributors. For application assistance,current specifications, pricing or thenearest Authorized Distributor, visit
sensing.honeywell.com or call:
Asia Pacific+65 6355-2828

Europe 		+44 1698 481481

USA/Canada+1-302-613-4491

Solstice® N40 is a registered trademark of Honeywell
International Inc.

HoneywellSensing and Internet of Things
9680 Old Bailes Road
Fort Mill, SC 29707
www. honeywell.com

32346591-B-EN | B | 01/20© 2020 Honeywell International Inc. All rights reserved.

m WARNINGPERSONAL INJURY
DO NOT USE these products as safetyor emergency stop devices or in anyother application where failure of the
product could result in personal injury.
Failure to comply with these
instructions could result in death orserious injury.