TEST

Oppdragsgiver Client

Utførende enhet/lab. Department/laboratory responsible

Nortek AS
Vangkroken 21351 Rud

Teknologisk Institutt as
Environmental Test Laboratory
P.O. Box 1019
NO - 3601 KONGSBERG

Rapportnr. Report no.

3000-16-024152
Tittel Title

Environmental Test Report
Vibration test
Signature 250

Dato Date

Utarbeidet av Prepared by

Godkjent av Approved by

Erik Haugejorden

14.06.2016

Innleveringsdato for prøve
Date of receipt of test object

13.06.2016
Prøvetaking utført av TI
Sampling by TI

Joachim Ski Torp
Revisjonsnr. Revision no.

Konfig.kont. Config.contr.

-Technical Manager
Antall sider No. of pages

Ant. vedlegg No. of append.

9Ja Yes

Kopi nr. Copy no.

Nei NoAkkreditert test (ISO 17025)
Accredited test (ISO 17025)

Kundens ref. Client’s ref.

Bengt Large

Ja Yes2Bestillingsnr. Order no.

Vibrasjon Signature 250

Fordeling Distribution

Bengt Large, Nortek; 1 pdf-file.

Akkreditering innebærer at laboratoriet oppfyller de krav Norsk
Akkreditering stiller til kompetanse og prøvingssystemer for de prøvinger
akkrediteringen omfatter. Dette innebærer også at laboratoriet har ettilfredsstillende kvalitetssikringssystem og nødvendig sporbarhet til
akkrediterte eller nasjonale laboratorier. Prøvingsresultatene gjelder ute
lukkende de prøvede objekter. Utdrag av rapporten må ikke gjengis uten
skriftlig godkjenning fra Teknologisk Institutt as.

SK-0016 d – N/E

Postadresse:
Pb. 141 Økern0509 Oslo
Besøksadresse:
Kabelgaten 2, 0580 Oslo

Accreditation implies that the laboratory meets the requirements set
by Norwegian Accreditation concerning competence and testing system
for all the testing contained in the accreditation. It also states that thelaboratory has a satisfactory quality assurance system and traceability
to accredited or national laboratories. The results relate only to theitems tested. The report shall not be reproduced except in full, without
the written approval of the institute.

Teknologisk Institutt as
Telefon:Telefaks:
E-post:Web:

22 86 50 00
22 72 45 02
firmapost@ti.no
www.teknologisk.no

Org.nr. 942 340 680

Kongsberg:
Stavanger:
Haugesund:Bergen:Pb. 1019, NO-3601 Kongsberg
Forusbeen 10, NO-4033 Stavanger
Pb. 93, NO-4299 Avaldsnes
Pb. 23, NO-5346 Ågotnes

Rapportnr. Report no.

3000-16-024152
Revisjonsnr Rev.nr. Side Page

2 av of 9
Dato Date 14.06.2016

Contents
1.
2.
3.
4.
5.

Scope .......................................................................................................................................................... 3
Reference Documents ................................................................................................................................ 3
Test Procedure............................................................................................................................................ 3
Results ........................................................................................................................................................ 4Conclusion ................................................................................................................................................. 4

Appendix 1: Test Equipment.
Appendix 2: Test Records.

Rapportnr. Report no.

3000-16-024152
Revisjonsnr Rev.nr. Side Page

3 av of 9
Dato Date 14.06.2016

1. Scope
The Environmental Test Laboratory at Teknologisk Institutt as has performed vibration test on Signature 250
S/N: D-0214 for Nortek AS. The test was performed 13.06.2016 and 14.06.2016.
The test was executed under accreditation granted by Norwegian Accreditation, accreditation no. Test 006,
and was based on the general test requirements specified in IEC 60068-1 / IEC 60068-2-64.
The following personnel were involved in the testing:Erik Haugejorden, Teknologisk Institutt AS.
Joachim Ski Torp, Teknologisk Institutt AS.
Bengt Large, Nortek AS.

2. Reference Documents
1.
2.
3.
4.

KS Handbook, Quality Manual for Teknologisk Institutt.
Environmental test report: 3010-09-0048, Teknologisk Institutt AS.
IEC 60068-1 General requirements, edition 7.0. 2013-10
IEC 60068-2-64 Test Fh: Vibration, broad band random and guidance, edition 2.0. 2008-04

3. Test Procedure
The environmental test equipment used during test with calibration status and measurement uncertainty
figures is given in Appendix 1. The test item were mounted on the shaker as shown in figures 1, 4 and 7. The
test item were not powered during test. A visual inspection was performed after test. Functional test wasperformed at Nortek after test. The positions of control and response accelerometers are shown in figures 2,3, 5, 6, 8 and 9.
MP1: Control accelerometer.
MP2: Response accelerometer on test item.
The following test parameters were used:
Frequency range:
10 - 500Hz.
Excitation level:
10 - 200 Hz: 0.03 g2/Hz.
200 - 500 Hz: 0.01 g2/Hz.
Grms:
2.95
Duration:
30 minutes in each of the 3 axes.
The recorded input and response records are given in Appendix 2.

Rapportnr. Report no.

3000-16-024152
Revisjonsnr Rev.nr. Side Page

4 av of 9
Dato Date 14.06.2016

4. Results
The test items were not operational during test. Mechanical inspection and functional tests were performed at
Nortek after the test. Test results are evaluated by Nortek.

5. Conclusion
The test items were not operational during test. Mechanical inspection and functional tests were performed at
Nortek after the test. Test results are evaluated by Nortek.

Rapportnr. Report no.

3000-16-024152
Revisjonsnr Rev.nr. Side Page

5 av of 9
Dato Date 14.06.2016

Figure 1 - Test item placed in X-axis

MP2

Figure 2 - MP2 in X-axis

Rapportnr. Report no.

3000-16-024152
Revisjonsnr Rev.nr. Side Page

6 av of 9
Dato Date 14.06.2016

Control acc.

Figure 3 - Control accelerometer in X-axis

Figure 4 - Test item placed in Y-axis

Rapportnr. Report no.

3000-16-024152
Revisjonsnr Rev.nr. Side Page

7 av of 9
Dato Date 14.06.2016

MP2

Figure 5 - MP2 in Y-axis

Control acc.

Figure 6 - Control accelerometer in Y-axis

Rapportnr. Report no.

3000-16-024152
Revisjonsnr Rev.nr. Side Page

8 av of 9
Dato Date 14.06.2016

Figure 7 - Test item placed in Z-axis

MP2

Figure 8 - MP2 in Z-axis

Rapportnr. Report no.

3000-16-024152Revisjonsnr Rev.nr. Side Page

9 av of 9
Dato Date 14.06.2016

Control acc.

Figure 9 - Control accelerometer in Z-axis

Rapportnr. Report no.

3000-16-024152
Revisjonsnr Rev.nr. Vedlegg Appendix
Side Page 1 av of
Dato Date

Appendix 1
Test Equipment

1

2
14.06.201622

Rapportnr. Report no.

3000-16-024152
Revisjonsnr Rev.nr. Vedlegg Appendix

1

Side Page 2 av of
Dato Date

Vibrator
Int.no.0883

214.06.20162
2

Vibrator
AmplifierInt.no TI126424 Vibration control sys
Int.no TI126423 Vibration control sys

LDS 964 LSLE 4014 ESpider 80 S
Spider 80 M

Cal.: N.A.
Cal.: N.A.
Cal.: 18.01.16
Cal.: 18.01.16

Interval: 12 months
Interval: 12 months

Int.no.02017

VibratorAmplifier
Int.no TI126422 Vibration control sys

LDS V810
LDS
Spider 80 M

Cal.: N.A
Cal.: N.A
Cal.: 15.12.15

Interval: 12 months

Int.no. 440495
Int.no. 440491
Int.no. 175417
Int.no. 171852

2079297
59979
C196900
C141500

Cal.: 17.12.15
Cal.: 26.11.15
Cal.: 07.04.16
Cal.: 21.09.15

Interval: 12 months
Interval: 12 months
Interval: 12 months
Interval: 12 months

Accelerometer
Accelerometer
Accelerometer
Accelerometer

Vibration fixtures were supplied by Nortek.
Table 1: Measurement uncertainty figures.
Parameter
Uncertainty
Acceleration+/- 2,95 %Coverfactor k2

Remarks

Figures based on specification andfigures from calibration of individual
items in measurement chain.
Frequency+/- 0.01 %2
Figures based on specification.
* Details from measurement uncertainty calculations can be forwarded on request.

Rapportnr. Report no.

3000-16-024152
Revisjonsnr Rev.nr. Vedlegg Appendix
Side Page 1 av ofDato DateAppendix 2
Test Records2414.06.201634

Rapportnr. Report no.

3000-16-024152
Revisjonsnr Rev.nr. Vedlegg Appendix
Side Page 2 av of
Dato Date

Run 1 – X-axis

MP22414.06.201634

Rapportnr. Report no.

3000-16-024152
Revisjonsnr Rev.nr. Vedlegg Appendix
Side Page 3 av of
Dato Date

Run 2 – Y-axis

MP224
14.06.201634

Rapportnr. Report no.

3000-16-024152
Revisjonsnr Rev.nr. Vedlegg Appendix
Side Page 4 av of
Dato Date

Run 1 – Z-axis

MP224
14.06.201634