DPR SERIES

32347489
Issue B

TruStability™ Board Mount Pressure Sensors
Standard Accuracy, Compensated/Ampliﬁed±1.6 mbar to ±25 mbar | ±160 Pa to ±2.5 kPa |±0.5 H2O to ±10 inH2O
DESCRIPTIONThe TruStability™ DPR Series is a

POTENTIAL INDUSTRIAL
APPLICATIONS

piezoresistive silicon pressure sensor• Heating, ventilation, air conditioning

offering a ratiometric analog output for(HVAC)

reading pressure over the specified full

• Variable air volume (VAV) control

scale pressure span and temperature• Damper control and duct air flow

range. These sensors are fully calibrated• Filter monitoring/clogged filter

and temperature compensated for

detection

sensor offset, sensitivity, temperature• Modulated furnace controls

effects, and non-linearity using an on-• Indoor air qualityboard Application Specific Integrated• Leak detection

Circuit (ASIC). Calibrated output• Pneumatic control

values for pressure are updated at• Burner control

approximately 1 kHz. The DPR Series is• Fuel-to-air rationing

calibrated over the temperature range• Gas analyzers and metersof -20°C to 70°C [-4°F to 185°F].

• Fume hoods and clean rooms

The sensor is calibrated for operationfrom a single power supply of 5.0 Vdc.

FEATURES

These sensors measure differential• Pressure range: ±1.6 mbar to ±25 mbar |or gage pressures. The robust and±160 Pa to ±2.5 kPa | ±0.5 inH2O todurable housing, with its industry-±10 inH2O

standard mounting configuration,• Pressure types: Differential and gagealong with a choice of flexible electrical• Total Error Band: As low as ±1.25 %FSS

terminations, is designed to withstanddepending on pressure range (after

tough application environments. The

auto zero)

sensors’ internal diagnostic functions• Accuracy: ±0.25 %FSS BFSL (Fulldetect most internal failures, including

Scale Span Best Fit Straight Line)burst sensors, and may reduce the• Stable offset voltage

need for redundant sensors in the• Compensated temperature range:system. The DPR Series is intended-20ºC to 70ºC [-4ºF to 158ºF]

for use with non-corrosive, non-

• Low sensitivity to mounting orientation

ionic gases, such as air and other dry• Output: analog voltage

gases. Custom calibrations, additional• Overvoltage and reverse polarity

pressure ranges or options to extend theperformance of these sensors to include
non-corrosive, non-ionic liquids isprotected• Industry-standard mounting
configuration and barbed ports

possible. All products are designed and• Robust and durable package

manufactured according to ISO 9001

• Internal diagnostic functions

standards.

• RoHS compliant

PORTFOLIO
Honeywell offers a variety ofboard mount pressure sensors forpotential use in medical and industrial
applications. Our categories ofpressure sensor measurement include
absolute, differential, gage or vacuum
gage, with unamplified or amplified
sensors and covering a wide pressure
range. To view the entire productportfolio, click here.

TRUSTABILITY™ BOARD MOUNT PRESSURE SENSORS, DPR SERIESTABLE 1. ABSOLUTE MAXIMUM RATINGS1
CHARACTERISTIC

MIN.

MAX.

UNIT

Supply voltage (Vsupply)

-5.0

12.0

Vdc

Voltage on output pin

-0.3

6.0

Vdc

1

Absolute maximum ratings are the extreme limits the device will withstand without damage.

TABLE 2. ENVIRONMENTAL SPECIFICATIONSCHARACTERISTIC

PARAMETER

Humidity

0 %RH to 95 %RH, non-condensing

Vibration

15 g, 10 Hz to 2 kHz

Shock

100 g, 6 ms duration

ESD susceptibility

3 kV min, human body modelLife

1 million pressure cycles minimum

11Life may vary depending on the specific application in which the sensor is utilized.

TABLE 3. WETTED MATERIALS1
MATERIALCOMPONENTPORT 1 (PRESSURE PORT)

Ports and covers

high temperature polyamide

Substrate

alumina ceramic

Adhesives

epoxy, silicone

O-Ringsilicone rubber

Electronic components
1

PORT 2 (REFERENCE PORT)

ceramic, silicon, glass, solder

silicon, glass, gold

Contact Honeywell Customer Service for detailed material information.

TABLE 4. SENSOR PRESSURE TYPESPRESSURE TYPEDESCRIPTION

Differential

Output is proportional to the difference between the pressures applied to each port (Port 1 - Port 2).

Gage

Output is proportional to the difference between applied pressure and atmospheric (ambient) pressure.

FIGURE 1. TEB COMPONENTS FOR TRUSTABILITY™
BOARD MOUNT PRESSURE SENSORSFIGURE 2. TRANSFER FUNCTION LIMITS

Sources of Error

Pressure Hysteresis
Pressure Non-Repeatability
Thermal Effect on Offset
Thermal Effect on Span
Thermal Hysteresis

Accuracy
BFSL

Total
Error
Band

Output (%Vsupply)

OffsetFull Scale Span
Pressure Non-Linearity

Compensated Pressure Range

100
90
80
70
60
50
40
30
20
10
0

Ideal
2% Total
Error Band

0

1

Pmin.
Output (V) =2sensing.honeywell.com

2

3

4

5

6

7

8

Pressure (example unit)

9

10

Pmax.

0.75 x Vsupply x (Pressure
applied - Pmin.) + 0.05 x Vsupply

Pmax. - Pmin.

TRUSTABILITY™ BOARD MOUNT PRESSURE SENSORS, DPR SERIESTABLE 5. OPERATING SPECIFICATIONSCHARACTERISTIC

MIN.

TYP.

MAX.

UNIT

Supply voltage (Vsupply)1

4.75

5.0

5.25

Vdc

Minimum operating voltage

3.0

—

—

Vdc

Supply current

—

2.7

3.5

mA

-40 [-40]—

85 [185]°C [°F]-20 [-4]—

70 [158]°C [°F]

Startup time (power up to data ready)

—

—

5

ms

Response time

—

1

—

ms

Clipping limits:
upper
lower

—
2.5

—
—

97.5
—

%Vsupply

Accuracy4

—

—

±0.25

%FSS BFSL6

Output resolution

—

0.033

—

%FSS6

<40 mbar | 4 kPa | 20 inH2O

—

±0.1

—

%FSS6

<2.5 mbar | 250 Pa | 1 inH2O

—

±0.2

—

Operating temperature range2
Compensated temperature range

3

Orientation sensitivity (±1 g)5:

Ratiometricity of the sensor (the ability of the device output to scale to the supply voltage) is achieved within the specified
operating voltage.
2
Operating temperature range: The temperature range over which the sensor will produce an output proportional to pressure.
3
Compensated temperature range: The temperature range over which the sensor will produce an output proportional to pressure
within the specified performance limits.
4
Accuracy: The maximum deviation in output from a Best Fit Straight Line (BFSL) fitted to the output measured over the pressure
range. Includes all errors due to pressure non-linearity, pressure hysteresis, and non-repeatability.
5
Orientation sensitivity: The maximum change in offset of the sensor due to a change in position or orientation relative to Earth’s
gravitational field.
6
Full Scale Span (FSS): The algebraic difference between the output signal measured at the maximum (Pmax.) and minimum(Pmin.) limits of the pressure range. (See Figure 3 for ranges).
1

Sensing and Internet of Things3TRUSTABILITY™ BOARD MOUNT PRESSURE SENSORS, DPR SERIESTABLE 6. DPR SERIES AVAILABILITY1CATALOG LISTINGDESCRIPTIONDPRCAN0005NG0000A5DPR Series, standard accuracy, compensated/amplified, flying leads with 304,8 mm harness
length, no gel, 0 inH2O to 5 inH2O pressure range, analog output, 5% to 80% of Vsupply transfer
function, 5.0 Vdc supply voltage

DPRAAN0005NG0000A5DPR Series, standard accuracy, compensated/amplified, AMP 3-643814-3 connector with
304,8 mm harness length, no gel, 0 inH2O to 5 inH2O pressure range, analog output, 5% to 80%
of Vsupply transfer function, 5.0 Vdc supply voltage

DPRCAN0002NG0000A5DPR Series, standard accuracy, compensated/amplified, flying leads with 304,8 mm harness
length, no gel, 0 inH2O to 2 inH2O pressure range, analog output, 5% to 80% of Vsupply transfer
function, 5.0 Vdc supply voltage

DPRABN0005NG0000A5DPR Series, standard accuracy, compensated/amplified, AMP 3-643814-3 connector with
50,8 mm harness length, no gel, 0 inH2O to 5 inH2O pressure range, analog output, 5% to 80%
of Vsupply transfer function, 5.0 Vdc supply voltage

DPRCAN0010NG0000A5DPR Series, standard accuracy, compensated/amplified, flying leads with 304,8 mm harness
length, no gel, 0 inH2O to 10 inH2O pressure range, analog output, 5% to 80% of Vsupply transfer
function, 5.0 Vdc supply voltage

These catalog listings are high volume and may be shipped quickly. Other configurations per Figure 3 are possible; however, minimum order
quantity thresholds and NRE may apply. Please consult the factory.

1

FIGURE 3. NOMENCLATURE AND ORDER GUIDEFor example, DPRCAN0002NG0000A5 defines a DPR Series, Standard Accuracy, Compensated/Amplified, flying leads, 304,8 mm
harness length, no gel, 0 inH2O to 2 inH2O pressure range, analog output type, 5% to 80% of Vsupply transfer function, 5.0 Vdc
supply voltage.

Product Series

D P R

accuracy,
DPR Standard
compensated/amplified

C

A

N

0 0 0 2 N G 0 0 0 0
Pmax.

unit,
reference5
Supply Voltage
5 5.0 Vdc

Electrical Termination

Transfer Function

AMP 3-643814-3

A

A

Pmin.

A 5% to 80% of Vsupply (analog output)
B 10% to 90% of Vsupply (analog output)

Pressure Range

Differential

AMP 1-480701-0

B

Flying leads

C
Molex 39-01-4033

D

Harness LengthA

304,8 mmB

50,8 mmGel
N No gel4

sensing.honeywell.com

01.6MDN1.6
02.5MDN2.5
0004MDN004
0006MDN006
0010MDN010
0016MDN016
0025MDN025±1.6 mbar

0160LDN160
0250LDN250
0400LDN400
0600LDN600
0001KDN001
01.6KDN1.6
02.5KDN2.5±160 Pa

00.5NDN0.5
0001NDN001
0002NDN002
0005NDN005
0010NDN010±0.5 inH20±2.5 mbar±4 mbar±6 mbar±10 mbar±16 mbar±25 mbar±250 Pa±400 Pa±600 Pa±1 kPa±1.6 kPa±2.5 kPa±1 inH20±2 inH20±5 inH20±10 inH20

Gage
02.5MG0000
0004MG0000
0006MG0000
0010MG0000
0016MG0000
0025MG0000
0040MG0000

0 mbar to 2.5 mbar

0250LG0000
0400LG0000
0600LG0000
0001KG0000
01.6KG0000
02.5KG0000
0004KG0000

0 Pa to 250 Pa

0001NG0000
0002NG0000
0005NG0000
0009NGN.75
0010NG0000
0014NGN.40
0020NG0000

0 mbar to 4 mbar
0 mbar to 6 mbar
0 mbar to 10 mbar
0 bar to 16 mbar
0 bar to 25 mbar
0 bar to 40 mbar

0 Pa to 400 Pa
0 Pa to 600 Pa
0 kPa to 1 kPa
0 kPa to 1.6 kPa
0 kPa to 2.5 kPa
0 kPa to 4 kPa
0 inH20 to 1 inH20
0 inH20 to 2 inH20
0 inH20 to 5 inH20-0.75 inH20 to 9 inH20
0 inH20 to 10 inH20-0.40 inH20 to 14 inH20
0 inH20 to 20 inH20

TRUSTABILITY™ BOARD MOUNT PRESSURE SENSORS, DPR SERIESTABLE 7. PRESSURE RANGE SPECIFICATIONS FOR ±1.6 MBAR TO ±25 MBAR DIFFERENTIAL AND 0 MBAR TO 2.5 BAR TO
0 MBAR TO 40 MBAR GAGE.
WORKINGOVER
BURST
PRESSURE1 PRESSURE2 PRESSURE3

COMMON
MODE
PRESSURE4

TOTAL
ERROR
BAND
(%FSS)5

Pmax.

PRESSURE UNIT
RANGE
Pmin.

PRESSURE
RANGE(SEE
FIGURE 3.)

TOTAL
LONG-TERMERRORSTABILITY,
BAND AFTER
1000 HR
AUTO-ZERO625°C(%FSS)(%FSS)

Differential
01.6MDN1.6

-1.6

1.6

mbar

335

675

1000

3450

±3.5%

±2.5%

±0.5%

02.5MDN2.5-2.5 2.5

mbar

335

675

1000

3450

±3%

±1.5%

±0.35%

0004MDN004

-4

4

mbar

335

675

1000

3450

±2.5%

±1.25%±0.35%

0006MDN006

-6

6

mbar

335

675

1000

3450

±2%

±1.25%±0.35%

0010MDN010

-10

10

mbar

375

750

1250

5450

±2%

±1.25%±0.25%

0016MDN016

-16

16

mbar

375

750

1250

5450

±2%

±1.25%±0.25%

0025MDN025

-25

25

mbar

435

850

1350

10450

±2%

±1.25%±0.25%

Gage

1

02.5MG0000

0

2.5

mbar

335

675

1000

3450

±4%

±3%

±0.5%

0004MG0000

0

4

mbar

335

675

1000

3450

±3%

±1.5%

±0.5%

0006MG0000

0

6

mbar

335

675

1000

3450

±3%

±1.5%

±0.35%

0010MG0000

0

10

mbar

335

675

1000

3450

±2%

±1.25%±0.35%

0016MG0000

0

16

mbar

335

675

1000

3450

±2%

±1.25%±0.25%

0025MG0000

0

25

mbar

375

750

1250

5450

±2%

±1.25%±0.25%

0040MG0000

0

40

mbar

375

750

1250

5450

±2%

±1.25%±0.25%

Working Pressure: The maximum pressure that may be applied to any port of the sensor in continuous use. This pressure may
be outside the operating pressure range limits (Pmin. to Pmax.) in which case the sensor may not provide a valid output untilpressure is returned to within the operating pressure range. Tested to 1 million cycles, minimum.

2

Overpressure: The maximum pressure which may safely be applied to the product for it to remain in specification once pressure
is returned to the operating pressure range. Exposure to higher pressures may cause permanent damage to the product. Unlessotherwise specified, this applies to all available pressure ports at any temperature within the operating temperature range

3

Burst pressure: The maximum pressure that may be applied to any port of the product without causing escape of pressure media.
Product should not be expected to function after exposure to any pressure beyond the burst pressure.

4

Common mode pressure: The maximum pressure that can be applied simultaneously to both ports of a differential pressure
sensor without causing changes in specified performance.

5

Total Error Band: The maximum deviation from the ideal transfer function over the entire compensated temperature and pressure
range. Includes all errors due to offset, full scale span, pressure non-linearity, pressure hysteresis, repeatability, thermal effect on
offset, thermal effect on span, and thermal hysteresis (see Figure 1).

6

Total Error Band after Auto-Zero: The maximum deviation from the ideal transfer function over the entire compensated pressure
range at a constant temperature and supply voltage for a minimum of 24 hours after an auto-zero operation. Includes all errors
due to full scale span, pressure non-linearity, pressure hysteresis, and thermal effect on span.

Sensing and Internet of Things5TRUSTABILITY™ BOARD MOUNT PRESSURE SENSORS, DPR SERIESTABLE 8. PRESSURE RANGE SPECIFICATIONS FOR ±160 PA TO ±2.5 KPA DIFFERENTIAL AND 0 PA TO 250 PA TO
0 PA TO 4 KPA GAGE
WORKINGOVER
BURST
PRESSURE1 PRESSURE2 PRESSURE3

COMMON
MODE
PRESSURE4

TOTAL
ERROR
BAND
(%FSS)5

PMAX.

PRESSURE UNIT
RANGE
PMIN.

PRESSURE
RANGE(SEE
FIGURE 3.)

TOTAL
LONG-TERMERRORSTABILITY,
BAND AFTER
1000 HR
AUTO-ZERO625°C(%FSS)(%FSS)

Differential
0160LDN160-160 160

Pa

33500

67500

100000345000

±3.5%

±2.5%

±0.5%

0250LDN250-250 250

Pa

33500

67500

100000345000±3%±1.5%±0.35%

0400LDN400-400 400

Pa

33500

67500

100000345000±2.5%±1.25%±0.35%

0600LDN600-600 600

Pa

33500

67500

10000034500±2%±1.25%±0.35%

0001KDN001

-1

1

kPa

37.5

75

125

545

±2%

±1.25%±0.25%

01.6KDN1.6

-1.6

1.6

kPa

37.5

75

125

545

±2%

±1.25%±0.25%

02.5KDN2.5-2.5 2.5

kPa

43.5

85

135

1045

±2%

±1.25%±0.25%

Gage

1

0250LG0000

0

250

Pa

33500

67500

100000345000

±4%

±3%

±0.5%

0400LG0000

0

400

Pa

33500

67500

100000345000

±3%

±1.5%

±0.5%

0600LG0000

0

600

Pa

33500

67500

100000345000±3%±1.5%±0.35%

0001KG0000

0

1

kPa

33.5

67.5

100

345

±2%

±1.25%±0.35%

01.6KG0000

0

1.6

kPa

33.5

67.5

100

345

±2%

±1.25%±0.25%

02.5KG0000

0

2.5

kPa

37.5

75

125

545

±2%

±1.25%±0.25%

0004KG0000

0

4

kPa

37.5

75

125

545

±2%

±1.25%±0.25%

Working Pressure: The maximum pressure that may be applied to any port of the sensor in continuous use. This pressure may
be outside the operating pressure range limits (Pmin. to Pmax.) in which case the sensor may not provide a valid output untilpressure is returned to within the operating pressure range. Tested to 1 million cycles, minimum.

2

Overpressure: The maximum pressure which may safely be applied to the product for it to remain in specification once pressure
is returned to the operating pressure range. Exposure to higher pressures may cause permanent damage to the product. Unlessotherwise specified, this applies to all available pressure ports at any temperature within the operating temperature range

3

Burst pressure: The maximum pressure that may be applied to any port of the product without causing escape of pressure media.
Product should not be expected to function after exposure to any pressure beyond the burst pressure.

4

Common mode pressure: The maximum pressure that can be applied simultaneously to both ports of a differential pressure
sensor without causing changes in specified performance.

5

Total Error Band: The maximum deviation from the ideal transfer function over the entire compensated temperature and pressure
range. Includes all errors due to offset, full scale span, pressure non-linearity, pressure hysteresis, repeatability, thermal effect on
offset, thermal effect on span, and thermal hysteresis (see Figure 1).

6

Total Error Band after Auto-Zero: The maximum deviation from the ideal transfer function over the entire compensated pressure
range at a constant temperature and supply voltage for a minimum of 24 hours after an auto-zero operation. Includes all errors
due to full scale span, pressure non-linearity, pressure hysteresis, and thermal effect on span.

6

sensing.honeywell.com

TRUSTABILITY™ BOARD MOUNT PRESSURE SENSORS, DPR SERIESTABLE 9. PRESSURE RANGE SPECIFICATIONS FOR ±0.5 INH2O TO ±10 INH2O DIFFERENTIAL AND 0 ±0 INH2O TO
20 INH2O GAGE
WORKINGOVER
BURST
PRESSURE1 PRESSURE2 PRESSURE3

COMMON
MODE
PRESSURE4

TOTAL
ERROR
BAND
(%FSS)5

PMAX.

PRESSURE UNIT
RANGE
PMIN.

PRESSURE
RANGE(SEE
FIGURE 3.)

TOTAL
LONG-TERMERRORSTABILITY,
BAND AFTER
1000 HR
AUTO-ZERO625°C(%FSS)(%FSS)

Differential
00.5NDN0.5-0.5 0.5 inH2O

135

270

415

1400

±4%

±3%

±0.5%

0001NDN001

-1

1

inH2O

135

270

415

1400

±3%

±1.5%

±0.35%

0002NDN002

-2

2

inH2O

135

270

415

1400

±2%

±1.25%±0.35%

0005NDN005

-5

5

inH2O

150

300

500

2200

±2%

±1.25%±0.25%

0010NDN010

-10

10

inH2O

175

350

500

4200

±2%

±1.25%±0.25%

Gage
0001NG0000

0

1

inH2O

135

270

415

1400

±4%

±3%

±0.5%

0002NG0000

0

2

inH2O

135

270

415

1400

±3%

±1.5%

±0.35%

0005NG0000

0

5

inH2O

135

270

415

1400

±2%

±1.25%±0.25%

0009NGN.75

-0.75

9

inH2O

135

270

415

1400

±2%

±1.25%±0.25%

0010NG0000

0

10

inH2O

150

300

500

2200

±2%

±1.25%±0.25%-0.40 14

inH2O

150

300

500

2200

±2%

±1.25%±0.25%

20 inH2O

175

350

550

4200

±2%

±1.25%±0.25%

0014NGN.40
0020NG0000
10Working Pressure: The maximum pressure that may be applied to any port of the sensor in continuous use. This pressure may
be outside the operating pressure range limits (Pmin. to Pmax.) in which case the sensor may not provide a valid output untilpressure is returned to within the operating pressure range. Tested to 1 million cycles, minimum.

2

Overpressure: The maximum pressure which may safely be applied to the product for it to remain in specification once pressure
is returned to the operating pressure range. Exposure to higher pressures may cause permanent damage to the product. Unlessotherwise specified, this applies to all available pressure ports at any temperature within the operating temperature range

3

Burst pressure: The maximum pressure that may be applied to any port of the product without causing escape of pressure media.
Product should not be expected to function after exposure to any pressure beyond the burst pressure.

4

Common mode pressure: The maximum pressure that can be applied simultaneously to both ports of a differential pressure
sensor without causing changes in specified performance.

5

Total Error Band: The maximum deviation from the ideal transfer function over the entire compensated temperature and pressure
range. Includes all errors due to offset, full scale span, pressure non-linearity, pressure hysteresis, repeatability, thermal effect on
offset, thermal effect on span, and thermal hysteresis (see Figure 1).

6

Total Error Band after Auto-Zero: The maximum deviation from the ideal transfer function over the entire compensated pressure
range at a constant temperature and supply voltage for a minimum of 24 hours after an auto-zero operation. Includes all errors
due to full scale span, pressure non-linearity, pressure hysteresis, and thermal effect on span.

Sensing and Internet of Things

7TRUSTABILITY™ BOARD MOUNT PRESSURE SENSORS, DPR SERIESFIGURE 4. NOMENCLATURE AND ORDER GUIDE
Harness Length Option“A” = 304,8 mm“B” = 50,8 mm

ø3,80[0.150]10,20[0.402]

Electrical Termination “A”:
AMP 3-643814-3 Pin 1

3,30[0.130]1,8[0.07]

ø3,18[0.125]

Pin 1: Red
Pin 2: BlackPin 3: Greenø5,21 ø4,57[0.205] [0.180]
Detail “A”

Electrical Termination “B”:
AMP 1-480701-0
Pin 1
Pin 1: Red
Pin 2: BlackPin 3: Green
12,70[0.500]3,80[0.150]Port 2 / Lo (-)

See detail “A”

Electrical Termination “C”:
Flying leads
Red wire: VSUPPLYPort 2 / Lo (-)

12,7[0.50]

Green wire: GroundBlack: Output +10,0[0.39]
31,2[1.23]

16,7[0.66]

Electrical Termination “D”:
MOLEX 39-01-4033 Pin 1

1,50[0.059]6,5[0.26]
Port 1

Port 133,38[1.314]

Pin 1: Red
Pin 2: BlackPin 3: Green

Catalog listing

55,90[2.201]

Wire Color Description
red
black
green

8

VSUPPLY
output +groundsensing.honeywell.com

Honeywell logo
Year
Month
Date
Lot

ø2,44
[0.096]ADDITIONAL MATERIALSWARRANTY/REMEDY

The following associated literature isavailable at sensing.honeywell.com:

Honeywell warrants goods of its
manufacture as being free of defective
materials and faulty workmanshipduring the applicable warranty period.
Honeywell’s standard product warranty
applies unless agreed to otherwise by
Honeywell in writing; please refer toyour order acknowledgment or consult
your local sales office for specificwarranty details. If warranted goods
are returned to Honeywell during theperiod of coverage, Honeywell will
repair or replace, at its option, withoutcharge those items that Honeywell,in its sole discretion, finds defective.
The foregoing is buyer’s sole remedy
and is in lieu of all other warranties,
expressed or implied, including thoseof merchantability and fitness for aparticular purpose. In no event shall
Honeywell be liable for consequential,
special, or indirect damages.

• Product range guide• Installation instructions• CAD Models

While Honeywell may provide
application assistance personally,
through our literature and the
Honeywell web site, it is buyer’ssole responsibility to determine
the suitability of the product in theapplication.

FOR MORE INFORMATION

Specifications may change without
notice. The information we supply is
believed to be accurate and reliable
as of this writing. However, Honeywellassumes no responsibility for its use.

Honeywell Sensing and Internet ofThings services its customers througha worldwide network of sales offices
and distributors. For application assistance, current specifications, pricing or
the nearest Authorized Distributor, visit
sensing.honeywell.com or call:
Asia Pacific+65 6355-2828

Europe 		+44 1698 481481

USA/Canada+1-302-613-4491

Honeywell Sensing and Internet ofThings9680 Old Bailes Road
Fort Mill, SC 29707
www. honeywell.com

32347489-B-EN | B | 01/20© 2020 Honeywell International Inc. All rights reserved.

m WARNINGPERSONAL INJURY
DO NOT USE these products as safetyor emergency stop devices or in any
other application where failure of the
product could result in personal injury.
Failure to comply with these
instructions could result in death orserious injury.

m WARNINGMISUSE OF
DOCUMENTATION••

The information presented in thisproduct sheet is for reference
only. Do not use this document as
a product installation guide.
Complete installation, operation,and maintenance information
is provided in the instructions
supplied with each product.

Failure to comply with these
instructions could result in death orserious injury.