DP2000G

WIDE RANGE
GRAPHIC PROGRAM CONTROLLER
DP2000G series is a graphic program controller
specially designed for wide range control. This
control unit divides a control domain into a highregion and a low region and controls extensively
by changing two sensors for each region at a set
point automatically.
Good accuracy and stability are realized for both
high and low temperature control by using an
optimum sensor for each region.
Automatic and Bumpless function realizes
smooth control performance on switching the ■ MODELSD P2 1 □□G □□□−□□□
regions.
INPUT
1 : Thermocouple/mV/V/mA
Control mode (output No.1)
1 : ON-OFF pulse PID
2 : ON-OFF servo PID (standard load spec)
3 : Current output PID (general type 4 to 20mA DC)
5 : SSR drive pulse PID
6 : Voltage output PID (general type 0 to 10V DC)
8 : ON-OFF servo PID (minimal load spec)
A : Current output PID (high accuracy type 4 to 20mA DC)
B : Current output PID (high accuracy type 1 to 5mA DC)
C : Voltage output PID (high accuracy type 0 to 10V DC)
Control mode (output No.2)
0 : None
1 : ON-OFF pulse PID *1 OP
3 : Current output PID(general type 4 to 20mA DC) *1 OP
5 : SSR drive pulse PID *1 OP
6 : Voltage output PID(general type 0 to 10V DC) *1 OP
A : Current output PID(high accuracy type 4 to 20mA DC) *1 OP
B : Current output PID(high accuracy type 1 to 5mA DC) *1 OP
C : Voltage output PID(high accuracy type 0 to 10V DC) *1 OP
Communications interface
0 : None
R : RS232C (COM1) OP
S : RS485 (COM1) OP
A : RS422A (COM1) OP
B : RS232C (COM1) + RS232C (COM2) OP
C : RS485 (COM1) + RS232C (COM2) OP
D : RS422A (COM1) + RS232C (COM2) OP
E : RS232C (COM1) + RS485 (COM2) OP
F : RS485 (COM1) + RS485 (COM2) OP
G : RS422A (COM1) + RS485 (COM2) OP
Transmission signal output 1
0 : None
1 : 4 to 20mA OP
2 : 0 to 1V OP
3 : 0 to 10V OP
4 : 1 to 5V OP
Transmission signal output 2
0 : None *2
1 : 4 to 20mA OP
2 : 0 to 1V OP
3 : 0 to 10V OP(Adding only transmission signal output 2 is not available)
Case color
G : Gray
B : BlackOP
External input/ output signal
0 : None OP
1 : Digital input/ output (non voltage contact input)
2 : Digital input/ output(External power supply spec for only input) OP
Transmitter power supply
0 : None
1 : Transmitter power supply OP■ FEATURES●Realizing wide range control by changing two sensors
for high and low regions
Two sensors can be connected for each high and low
region sensing. By switching to an appropriate sensor foreach region, continuous control is available. Various
changing types are selectable such as automatic change at
a set point, changing by external input signal and the
combination of both.
●Employing clear 5.6”TFT color LCD display
Graphic screen of pattern progress status, display PV
value/SV value/pattern/step/time and various monitor
function such as trend display, enlarged data display andbar graph display are prepared.
●Easy program pattern setting on graphic screen
Maximum 200 patterns, 199 steps/patterns (total 4000
steps) settings are available as well as pattern repeat,
linking between patterns and endless program setting.
●Parameter settings per step
Each parameter setting such as PID constant, alarm,
guarantee soak and time signal is available per step.
●High performance and universal input
Input selection from each range of thermocouple, DC
voltage/current are available and unit has performance of 5
digits display, accuracy rating of ±0.1% and sampling
period of 0.1sec.
●Storing settings in CF card
Setting management is easy as all settings including setting
program pattern and each parameter are stored in CF cardand readout from it. PC software allows you to edit program
pattern and parameter.
●Various control application functions
2-output specification such as Heating/cooling output isapplicable as control output has ON-OFF pulse type,
ON/OFF servo type, current output type, SSR drive pulse
type and voltage output type.
●Abundant external input/output
Unit with external input 16 points and external output 28
points enables function assignment. Synchronized
operation with peripherals is easy. Serial communications
interface and transmission signal output are also prepared.
●Interchangeable with DP series
DP series are easily replaced with DP2000G which has
inherited characteristics of the DP series such as function,
operability and terminal arrangement/configuration.

*1 Selectable when control mode (output1) is 1,3,5,6,A,B,C.
*2 COM1 is exclusive use for rear port,
COM2 is rear port and front port switchable.
OP Option1

PSE-809A

DP2000G■OPERATION SCREEN●Running status display at once
Running status display of pattern progress and PV/ SV/
MV/ variation.

●Trend screen
Enlarged trend display of PV and SV●Enlarged data screen
Enlarged display of PV/SV●Bar-graph screen
Bar-graph display of PV/ SV/ MV●Pattern setting screen●Step parameter setting screen■KEY ARRANGEMENT

A/M

RUN

RESET

DISP

ESC
Direction key

FNC

STOP

ADV

MENU

ENT

2

Auto tuning:

AT1 --- Set by the target value during operation
AT2 --- Preset the step interval coaxial 8 types
AT3 --- Preset 8 automatic selection types for SV interval
AT4 to AT6 --- Setting for the 2 outputs typeOn-off pulse type: Output signal --- On-off pulse conductive signal (relay
contact)
Contact capacity --- Resistance load 100 to 240VAC
30VDC, 5A or lessInductive load 100 to 240V AC 30 VDC,
2.5A or less
Minimum load 5 VDC, 10mA or more
Contact protection---CR element built-in
On-off servo type: Output signal --- On-off servo conductive signal
Contact capacity--- Standard load spec
Resistance load --- 100 to 240VAC
30VDC 5A or lessInductive load --- 100 to 240VAC
30VDC 2.5A or less
Minimum load --- 5VDC, 10mA or more
Minimal load spec
Resistance load --- 100 to 240 VAC
30VDC 20mA or lessInductive load --- 100 to 240 VAC
30VDC 20mA or less
Minimum load --- 5VDC, 1mA or more
Feedback resistance --- 100Ω to 2kΩ
Contact protection --- Compact CR
element built-in
Current output type:Output signal --- 4 to 20mA or 1 to 5mA
Load resistance --- 750Ω or lessControl output accuracy --- 0.1% for high accuracy type
SSR drive pulse type:Output signal --- On-off pulse voltage signal
At ON --- 12VDC±20% (maximum 20mA)
At OFF --- 0.8VDC or less
Voltage output type:Output signal --- 0 to 10 VDC
Output resistance --- Approx. 10Ω
Control output accuracy --- High accuracy type 0.1%
Output limit:
Upper 0.0 to 105.0%, Lower −5.0 to 100.0%
Output variation limit:Up 0.01 to 100.00%
Down −0.01 to −100.00%
Output preset:
Output setting in proportional operation when PV=SV−100.0% to 100.00%
Output dead band: Dead band setting 0.0 to 9.9%
(0.1 to 9.9% for 2 position operation)
Control action:
Direct/ reverse action switching
Guarantee soak: Deviation setting 0 to 99999, decimal point linked withscaling
Output at PV error:Individual setting of outputs at upper and lower limit errors−5.0 to 105.0%
A.R.W:
Upper 0.0 to 100.0%, lower −100.0 to 0.0%
Constant value operation:
Program (PROG) / constant (CONST) modeswitching
Manual operation: Output range --- −5.0 to 105.0%
・Balanceless bumpless when switching from MAN to AUTO
・Output at AUTO kept when switching from AUTO to MAN
Program actions on repower:
Select to continue or reset the program when recovering thepowerControl operation: Position type and speed type selectable
2 outputs specification:Independent PID, Any combination of 6 types from On-off
pulse type, current output type, SSR drive type, voltage
output type, current output type (high accuracy), voltage
output type (high accuracy)(No secondary output for ON-OFF servo type)Heating and cooling control:
Cooling proportional operation, matching box operation
Cascade primary controller:
Output (%) =a x control operation value + b + c x set valuea, c: 0.00 to 1.00, b: −100.0 to 100.0
Output destination − control output 1/2,
transmission output 1/2■ INPUT SPECIFICATIONS
Input points:
Input types:

2 points (Range-L /Range-H)
DC voltage --- ±10mV, ±20mV, ±50mV, ±100mV, ±5V,
±10V
DC current --- 20mA
Thermocouple --- B, R, S, K, E, J, T, N, U, L, WRe5-WRe26,
W-WRe26, NiMo-Ni, CR-AuFe, PR5-20,
PtRh40-PtRh20,Platinel
Accuracy rating: Refer to the table of measuring range and accuracy ratings
Reference junction compensation accuracy:
K, E, J, T, N, Platinel --- ±0.5℃ or less
Other than above --- ±1.0℃ or lessSensor bias:
One kind (bias) for Range-H, Liner correction of 9 breaking
points for Rang-L
Input change:
Input change by change SV (automatic), external input
signal or both.
Function: Bumpless (PV variation limit) at switching, PV
start, Change error status, Range-H status, Dead band
Sampling period: Approx. 0.1 sec
Burnout:
Burnout available for thermocouple, DC voltage (±50mV orless) and resistance thermometer
Output value at burnout is settable to any value
Range setting:
The useable range is settable within the measuring range(only for linear range)
Scaling:
DC voltage/ current input(Setting range: −99999 to 99999, decimal point specified)User linearize table:Useable for DC voltage/ current input (19 break points)
Digital filter:
0 to −99.9sec
Allowable signal source:
Thermocouple input/ DC voltage (mV) --- 100Ω or less
DC voltage input (±5V, ±10V) --- 300Ω or less
Resistance thermometer (3 wire) --- 5Ω or less per wire
(4 wire) --- 100Ω or less per wire
Input resistance: Thermocouple/DC voltage input --- 1MΩ or more
DC current input --- Approx. 100Ω
Measuring current: Resistance thermometer input --- Approx. 1mA
Maximum allowable input: Thermocouple/DC voltage input --- ±20V DC
DC current input --- ±30mA
Operation function:Square roots calculation, Log operation■ PROGRAMMING SPECIFICATIONSPattern set type:

Target temp (SV)/Time or Ramp rate/TimeTime setting - Hour/Minute or Minute/Second
Ramp rate setting - Temperature/minute or temperature/secondNumber of steps: Up to 199 steps per pattern
Number of patterns:Up to 200 patterns
Total number of steps:Up to 4000 steps
Repeat:Pattern --- Up to 9999 times, Step --- up to 99 times
Step setup range: Target value --- Input scale range
Ramp rate --- -99.999 to 99.999
Time --- 0 to 999 hours 59 minutes or 0 to 999 minutes 59
seconds
Start temperature: Select either PV start or arbitrary set value start
Target value (SV) correction:-99999 to 99999, decimal point linked with scaling
Fast-forward:
Program fast-forward function provided(FAST)(Approx. 10 times or 60 times)
End output:
Select either constant value control orfixed output(setting: -5 to 105%)Parameter registration:Each parameter is selectable per step
(Sequence programming) . PID constant --- 8 types, or 8 automatic selection types
for SV interval (including dead band, ARW upper/lower
limits, and output preset). Output limit (upper/lower)/ output variation limit(upper/lower) 8 types for each, or 8 automatic selection
types for SV interval. Guarantee soak 8 types. Wait time alarm 8 types. Alarm 8 types for each (a set of 4points). Time signal 30 types, all ON, all OFF, reverse phase,
repeat in a step
. Mass flow SV 8 types
Parameter setting change:
Changeable during operation
Target value, time, ramp rate, PID, ARW, guarantee soak,
output limit, output variation limit, alarm, SV, mass flow SV
Additional function:Pattern link, circle function, pattern edit

■ ALARM SPECIFICATIONS
Number of set points: 4 points + 4 points (for extended assignment setting)Judgment method: Upper alarm or lower alarm (with/without wait) using an
absolute value
Upper alarm or lower alarm (with/without wait) using andeviation
Upper alarm or lower alarm (with/without wait) using aabsolute value deviation
Upper alarm or lower alarm (with/without wait) using an
measured value change rate
Upper or lower limit judgment of output value (with/without
wait)
Upper or lower limit judgment of set value (with/without
wait)
Control loop error, fail, wait time alarm, end signal
Delay or latch function is selectable■ CONTROL SPECIFICATIONSControl switching period: Approx. 0.1 (initial value)/ 0.2/ 0.3/ 0.5 sec
Control type:
ON-OFF pulse type, ON-OFF servo type, current output
type, SSR drive pulse type, voltage output type
PID value:
Automatic setting by auto tuning or Manual setting
P --- 0 to 999.9% (0 for 2 position operation)
I --- 0 to 9999 sec (0 for no I operation)
D --- 0 to 9999 sec

3

DP2000G
Function:

Setting range:-99999 to 99999, decimal point linked with scaling
Dead band:
0.1 times of set resolution
Delay setting range:1 to 10 times
Output type:
Relay contact output 4 points --- (A contact, 1 common)
Contact capacity --- Resistance load 100 to 240VAC 30VDC,
3A or lessInductive load 100 to 240VAC 30VDC, 1.5A or less
External output signal assignment 4 points (for extended
assignment setting)
Alarm reset:
Alarm can be cleared during occurrence■ GENERAL SPECIFICATION
Rated power voltage:100 to 240V AC 50/60Hz (universal power supply)
Maximum power consumption: 50VA
Reference operation condition:
Ambient temperature humidity range --21 to 25℃, 50 to 60%RH
Power voltage --- 100V AC ±1.0%
Power frequency --- 50/60Hz ±0.5%
Attitude --- Left/right ±3°, forward/backward ±3°
Warm-up time --- 30 minutes or more
Normal operation condition:
Ambient temperature humidity range ---10 to 50℃, 10 to 90%RH
Power voltage --- 90 to 264V AC
Power frequency --- 50/60Hz ±2%
Attitude --- Left/right ±10°
, forward/backward ±10°Transportation condition:
At the packed condition on shipment from ourfactory
Ambient temperature humidity range ---20 to 60℃, 5 to 90%RH (No dew condensation)
Vibration --- 10 to 60Hz 0.5G (4.9m/s2) or less
Impact --- 40G (352m/s2) or less
Storage condition: Ambient temperature humidity range ---20 to 60℃, 5 to 90%RH(No dew condensation)
Power failure protection:
The settings are kept using EEPROM and lithium
battery backed up RAM
Insulation resistance:Between secondary terminal and protection conductor
terminal --- 500V DC 20MΩ or more
Between primary terminal and protection conductor
terminal --- 500V DC 20MΩ or more
Between primary terminal and secondary terminal---500V DC 20MΩ or more
Withstand voltage: Between secondary terminal and protection conductor
terminal --- 500V AC for 1minuteBetween primary terminal and protection conductor
terminal --- 1500V AC for 1minuteBetween primary terminal and secondary terminal---1500V AC for 1minute*Primary terminal: Power supply (100-240V AC), control
output terminals, and alarm output terminals*Secondary terminal: All terminals other than primaryterminal
Protection:
Conformed to IP54
Case assembly material:
Case, Front bezel, input/output terminal board--- Fire-retardant polycarbonate resin
External input/output , transmission output,
communications terminal board --- PBT
Color:
Front bezel, case --- Gray or black
Terminal cover:Standard provision
Weight:Approx 1.7kg
Mounting:
Panel mounting
Terminal screw:
M3.5 (M3 for external input/output, transmission output,
communications terminal board)■ EXTERNAL OUTPUT SIGNAL SPECIFICATION
Number of output: 28 points (function assignment per point)
Output type:Open collector output (24V DC, up to 50mA)
Time signal output:Default assignment --- 18 points
Output type --- ALL-ON/ ALL-OFF/ maximum of 30 types
per step
Status output:
Default assignment --- 10 points
Output type --- RUN/STOP, ADV, RESET, WAIT, FAST, END,
ALM-WAIT, ERR, SV-UP, SV-DOWN
Selective assignment --- Pattern/ step No.-BCD output
Alarm output:
Selective assignment --- 8 types
Output type --- AL1 to AL8■ EXTERNAL INPUT SIGNAL SPECIFICATION
Number of inputs: 16 points (function assignable per point except external
drive input)
Input type:
Non voltage contact (contact capacity 12V DC, 2mA or
more)
External power supply specification 12/24V DC ON whenpower is applied (up to 12mA/point)
External drive input:Default assignment --- 5 points
Input type --- RUN/STOP, ADV, RESET, WAIT, FAST
Selective assignment --- Circle pulse (program operation)
External A/M switching, alarm
reset, PV hold, SV holdPattern select input:Default assignment --- 10 points
Input type --- 10 types of 1, 2, 4, 8, 10, 20, 40, 80, 100, 200
Selection method --- Select the number from 1 to 200 using
BCD code■ DISPLAY SPECIFICATION
Screen:
Display content:

LCD backlight:

5.6"TFT color LCD
Operation screen
Home screen --- Pattern progress, pattern/step No.
numeric data, status, time signal, alarm
Enlarged data screen, bar-graph screen, trend screen,
DO/DI screen
Setting screen --- Pattern/sequence setting, various
parameter setting, memory card management setting,
maintenance, setting lock, communications, settingchange during operation
4 brightness adjustment levels■ SETTING AND OPERATION SPECIFICATION
Operation key type:
MENU, DISP, DIRECTION key, ENT, ESC, FNC, RUN,
STOP, ADV, RESET, A/M
Setting and operation method:
Setting --- Menu calling/ cursor selection method
Operation --- Direct key operation (combined with FNC)
Menu setting:Mode 0 (Execution steps setting)Mode 1 (Operation status selection/ Input change setting)Mode 2 (Pattern and sequence)
Mode 3 (PID/alarm)Mode 4 (Output/control)
Mode 5 (Input)
Mode 6 (Time signal/guarantee soak)
Mode 7 (Transmission)
Mode 8 (Communications)
Mode 9 (Memory card)
Mode 10 (Enhanced setup)
Mode 11 (Maintenance)
Mode 12 (Help)Operation:
Operation start/stop (RUN/STOP), operation reset (RESET),
Stepping operation (ADV), auto/manual switching(A/M), Fast-forwarding (FAST)
Display operation: Switching between operation screens
HOME screen (registered operation screen) automatic
displayEngineering port: Serial port on the front panel(Custom cable connection)■ SOFTWARE
DP-G parameter editing software.Program pattern editing / file management / printing.Setting parameter editing / file management / printing.CF card reading / storing for DP-G

■ OPTION SPECIFICATION●Transmission signal output
Number of outputs: Up to 2 points
Output signal:
4 to 20mA DC (load resistance 400Ω or less)0 to 1V DC (load resistance 50kΩ or more)
1 to 5V DC (load resistance 50kΩ or more)
0 to 10V DC (load resistance 50kΩ or more)*1 to 5V DC for secondary transmission output
Output accuracy: Primary output --- ±0.1% of output span
Secondary output --- ±0.3% of output span

●Transmitter power supply (Insulation type)
Power voltage:
24V DC
Current capacity: Up to 30mA

●Communications interface■ MEMORY CARD SPECIFICATION (Card is optional)
Memory media:
Memory size:
Saved data:

Save/read/delete/verify
For program patterns, individual or all pattern save/delete
selectable
Card format (simple format)

Number of communications points: Up to 2 points
Communications type:
RS232C, RS422A, RS485*COM2 for front and rear switching
Protocol:MODBUS/PRIVATE

Compact flash (CF) card
Up to 2 GB
Setup parameters, program patterns
All data (for auto loading)4

■ ACCURACY RATINGS■MEASURING RANGES
Measuring range
BRS

0.0 to 1820.0ºC

B

400 to 800ºC : ±0.2%±1digit

R、SN

0.0 to 1760.0ºC-200.0 to 1370.0ºC

K

0 to 400ºC : Not defined

0.0 to 1760.0ºC
0.0 to 1200.0ºC

0 to 400ºC : ±0.2%±1digit-200 to 0ºC: ±0.2%±1digitKor ±60μV-equivalent value, whichever is greater

0.0 to 600.0ºC-200.0 to 300.0ºC-270 to 0ºC: ±0.2%±1digitEor ±80μV-equivalent value, whichever is greater-270.0 to 1000.0ºC

E

0.0 to 700.0ºC-200.0 to 1200.0ºC

/

T C

J

WRe5-WRe26
W-WRe26
NiMo-Ni
CR-AuFeN
PtRh40-PtRh20
PlatinelDC voltage
DC current

U
L
10mV
20mV
50mV
100mV
5V
10V
20mA

/

-200 to 0ºC:±0.2%±1digit±0.1%±1digit

or ±40μV-equivalent value, whichever is greater-200 to 0ºC: ±0.2%±1digitUor ±40μV-equivalent value, whichever is greater-200.0 to 400.0ºC-270.0 to 400.0ºC-200.0 to 200.0ºC
0.0 to 2310.0ºC
0.0 to 2310.0ºC-50.0 to 1410.0ºC

L
WRe5-WRe26
W-WRe26
NiMo-Ni
Platinel-200 to 0ºC: ±0.2%±1digit
0 to 400ºC: ±0.3%±1digit

0 to 20K: ±0.5%±1digit

CR-AuFe

0.0 to 280.0K
0.0 to 1300.0ºC±0.2%±1digit

PtRh40-PtRh20
DC voltage / current

20 to 50K: ±0.3%±1digit
0 to 400ºC: ±1.5%±1digit
400 to 800ºC: ±0.8%±1digit

0.0 to 1880.0ºC
0.0 to 1390.0ºC

or ±80μV-equivalent value, whichever is greater-270 to 0ºC: ±0.2%±1digit

T

T C-200.0 to 900.0ºC-100.0 to 200.0ºC

TJ-270.0 to 300.0ºC-270.0 to 150.0ºC

Exception

Accuracy rating

Input type

Scale range±0.1%±1digit

0.0 to 600.0ºC-200.0 to 400.0ºC-200.0 to 900.0ºC-10 to 10mV
-20 to 20mV
-50 to 50mV
-100 to 100mV
-5 to

5 V

-10 to

10 V

0 to

20 mA

*Accuracy converted to the measuring range under the reference operation condition.
Reference junction compensation accuracy is added to thermocouple.
*K, E, J, T, R, S, B, N : IEC584 (1977, 1982), JIS C 1602-1995, JIS C 1605-1995,
2, CR-AuFe, PtRh40-PtRh20 : ASTM Vol.14.03 U, L : DIN43710-1985
WRe5-WRe26, W-WRe26, NiMo-Ni, Platinel2

5

■ TERMINAL ARRANGEMENT

DO terminals

100-240V AC

Control output terminals(Output1)

1

A

B C D

E F G H

I

J

2
3

Current/ voltage/ SSR drive pulse type (+)
M3
Current/ voltage/ SSR drive pulse type (-) On-off
servo type M2
M1
R1
RC
R2

H
C
L

On-offpulse type

16
1

17

6 3
7

18

8

19

9 4

20

10

21

11 5

22

StatusExternal drive/ pattern selection/ DI terminals
Thermocouple/DC voltage/DC current (+) terminal
Range-L terminals
Thermocouple/ DC voltage/ DC current (-) terminal

12
13 6

23

14

24

Thermocouple/DC voltage/DC current (+) terminal
Thermocouple/ DC voltage/ DC current (-) terminal

1

RS232C/RS485（COM2）
RS232C/RS485（COM1）
RS422A（COM1）

2
3

Control output terminals(Output2)

4

On-offpulse type

H
C
L

Output 2 (-) terminal
Output 2 (+) terminal
Output 1 (-) terminal
Output 1 (+) terminal

7
8
9

Range-H terminals

Communications interface
terminals (option)*RS422: COM1 port only
*COM1 is exclusive use for rear port
COM2 is rear port and front port switchable

5
6

Current/ voltage/
SSR drive pulse type (+)
Current/ voltage/
SSR drive pulse type (-)

Transmitter power supply terminals(option)Time signal

4 2
5

AL1
AL2
Alarm output terminals AL3
AL4
COM

Output (+) terminal
Output (-) terminal

15

Analog transmission output terminals (option)■ EXTERNAL INPUT/OUTPUT TERMINALSTime signal output terminals

Status output terminals

COM：1A ○TS 1 ：1B ○ ［LOAD］
TS 2 ：1C ○ ［LOAD］
TS 3 ：1D ○ ［LOAD］
TS 4 ：1E ○ ［LOAD］
TS 5 ：1F ○ ［LOAD］
TS 6 ：1G ○ ［LOAD］
TS 7 ：1H ○ ［LOAD］
TS 8 ：1I ○ ［LOAD］
TS 9 ：1J ○ ［LOAD］
COM：2A ○TS10：2B ○ ［LOAD］
TS11：2C ○ ［LOAD］
TS12：2D ○ ［LOAD］
TS13：2E ○ ［LOAD］
TS14：2F ○ ［LOAD］
TS15：2G ○ ［LOAD］
TS16：2H ○ ［LOAD］
TS17：2I ○ ［LOAD］
TS18：2J ○ ［LOAD］

COM ：3A ○RUN/STOP：3B ○ ［LOAD］
ADV：3C ○ ［LOAD］
RESET：3D ○ ［LOAD］
WAIT：3E ○ ［LOAD］
FAST：3F ○ ［LOAD］
END：3G ○ ［LOAD］
ALM・WAIT：3H ○ ［LOAD］
ERROR：3 I ○ ［LOAD］
SV・UP：3 J ○ ［LOAD］
SV・DOWN：4 J ○ ［LOAD］

External drive input terminals and pattern selection input (BCD code) terminals
External drive signal
COM ：4A ○RUN/STOP：4B ○ADV：4C ○RESET：4D ○WAIT：4E
FAST：4F ○
（BLK）
：4G ○
BCD code 100：4H ○200：4 I ○COM：5A ○1：5B ○
2：5C ○
4：5D ○
8：5E ○
10：5F ○
20：5G ○
40：5H ○
80：5 I ○*COM: Common to time signal output

COM ：4A ○RUN：4B ○STOP：4C ○RESET：4D ○ADV：4E ○
（BLK）
：4F ○
（BLK）
：4G ○
100：4H ○200：4 I ○12V/24V
COM：5A ○1：5B ○
2：5C ○
4：5D ○
8：5E ○
10：5F ○
20：5G ○
40：5H ○
80：5 I ○*External power supply spec
Applicable to equivalent of DP-I spec

■ DIMENSIONS

144
16

223
235
252

138 ＋01

170

170

159

144

137×137● PANEL CUTOUT

138 ＋01

Unit: mm

Specifications subject to change without notice. Printed in Japan (I) 2018. 8

32-8 KUMANO-CHO,ITABASHI-KU,TOKYO 173-8632
Telephone : +81-3-3956-2171
Facsimile : +81-3-3956-0915
E-mail : inter@chino.co.jp
Website : www.chino.co.jp/$