GP01191P/00/EN/01.22-00
71606665
2022-12-20

Products

Solutions

Valid as of version
01.00.zz (Device firmware)

Description of Device Parameters
Deltabar PMD55B
Differential pressure measurement
PROFINET with Ethernet-APL

advanced physical layer

ServicesDeltabar PMD55B PROFINET with Ethernet-APL

Table of contents

Table of contents
1

About this document . . . . . . . . . . . . . . . . 4

1.1
1.2
1.3
1.4
1.5

Document function . . . . . . . . . . . . . . . . . . . . . 4
Target group . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Using this document . . . . . . . . . . . . . . . . . . . . 4
Symbols used . . . . . . . . . . . . . . . . . . . . . . . . . . 5
Documentation . . . . . . . . . . . . . . . . . . . . . . . . 5

2Overview of the operating menu . . . . . 6

3

Description of device parameters . . . 19

3.1
3.2
3.3
3.4
3.5
3.6

User navigation . . . . . . . . . . . . . . . . . . . . . . . 19
"Device information" menu . . . . . . . . . . . . . . . 22
"Guidance" menu . . . . . . . . . . . . . . . . . . . . . . 23
"Diagnostics" menu . . . . . . . . . . . . . . . . . . . . . 38
"Application" menu . . . . . . . . . . . . . . . . . . . . . 54
"System" menu . . . . . . . . . . . . . . . . . . . . . . . . 72

Index . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 96

Endress+Hauser

3About this document

Deltabar PMD55B PROFINET with Ethernet-APL

1

About this document

1.1

Document function

The document is part of the Operating Instructions and serves as a reference forparameters. The document provides a detailed explanation of each individual parameter.
Performance of tasks that require detailed knowledge of the functioning of the device:• Commissioning measurements under difficult conditions• Optimal adaptation of the measurement to difficult conditions• Detailed configuration of the communication interface• Error diagnostics in difficult cases

1.2

Target group

The document is aimed at specialists who work with the device over the entire life cycle
and perform specific configurations.

1.3

Using this document

1.3.1

Information on the document structureThis document lists the submenus and parameters that are available when the"Maintenance" option user role is enabled.
For the operating concept of the operating menus, see the Operating Instructions.

1.3.2

Structure of a parameter description

The individual parts of a parameter description are described in the following section:• Navigation: Navigation path to the parameter via the local display• Prerequisite: The parameter is only available under these specific conditions• Description: Description of the parameter function• Selection: List of the individual options for the parameter• User entry: Input range for the parameter• User interface: Display value/data of the parameter• Additional information:• On individual options• On display values/data• On the input range• On the factory setting• On the parameter function

4

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

1.4

Symbols used

1.4.1

Symbols for certain types of Information

About this document

Additional information:
Reference to documentation:
Operation via local display:
Operation via operating tool:
Write-protected parameter:

1.5

Documentation

1.5.1

Standard documentation

Operating Instructions
The Operating Instructions are available via the Internet: www.endress.com →Download

1.5.2

Supplementary device-dependent documentation

Special Documentation
The Special Documentation is available via the Internet: www.endress.com →Download

Endress+Hauser

5Overview of the operating menu

Deltabar PMD55B PROFINET with Ethernet-APL

2

Overview of the operating menu

Navigation



Operating tool→  22

Device information
Status signal→  22Pressure→  22

Scaled variable→  22

Do not show this message again→  22→  19

Guidance‣ Commissioning→  23‣ Device identification
Device tag→  23

Device name→  23

Serial number→  23Extended order code 1→  24

Extended order code 2→  24

Extended order code 3→  24‣ Device identification
Locking status‣ Device identification→  25→  25→  26Time zone→  26

Date/time→  27‣ Device identification6→  23→  27

PROFINET device name→  27

PROFINET device name→  27

IP address→  27

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Overview of the operating menu

Descriptor→  28

MAC address→  28Device ID→  28Manufacturer ID→  28‣ Measurement adjustments→  29Damping→  29Assign scaled variable?

→  29‣ Measurement adjustments
Pressure unit

→  29Temperature unit

→  30‣ Measurement adjustments→  30Pressure unit

→  30Scaled variable unit

→  31

Free text→  31

Temperature unit

→  32‣ Measurement adjustments→  32

Zero adjustment→  32Pressure→  33‣ Output settings→  33
Scaled variable transfer function→  33Low flow cut off→  33‣ Output settings

Endress+Hauser→  29→  34Lower Range Limit→  34Upper Range Limit→  34

Minimum span

→  347

Overview of the operating menu

Deltabar PMD55B PROFINET with Ethernet-APL‣ Output settings→  34Pressure→  34

Scaled variable→  35‣ Output settings→  35
Scaled variable transfer function→  35Pressure value 1→  35

Scaled variable value 1→  36

Pressure value 2→  36

Scaled variable value 2→  36Lower Range Limit→  36Upper Range Limit→  36

Minimum span

→  37‣ Output settings
Assign process variable→  37

Assign process variable→  37

Assign process variable→  37

Assign process variable→  37

Assign process variable→  37

Assign process variable→  37

Assign process variable→  37→  38Diagnostics‣ Active diagnostics8→  37→  38

Active diagnostics→  38Timestamp→  38Previous diagnostics→  39Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Timestamp→  39Operating time from restart→  39Operating time→  40‣ Event logbook→  40Filter options‣ Minimum/maximum values→  40→  41Pressure min→  41Counter limit underruns sensor Pmin

→  41Counter underruns of user limit Pmin

→  41Minimum sensor temperature→  41Counter limit underruns sensor Tmin

→  42

Counter underruns of user limit Tmin

→  42

Minimum terminal voltage→  42

Minimum electronics temperature→  42

Reset user defined counters P and T→  42Pressure max→  43

Counter limit overruns sensor Pmax

→  43

Counter overruns of user limit Pmax

→  43

Maximum sensor temperature→  43

Counter limit overruns sensor Tmax

→  44

Counter overruns of user limit Tmax

→  44

Maximum terminal voltage→  44

Maximum electronics temperature→  44‣ Simulation→  45
Simulation

Endress+Hauser

Overview of the operating menu

→  459

Overview of the operating menu

Deltabar PMD55B PROFINET with Ethernet-APL

Diagnostic event simulation→  45

Value pressure simulation→  45‣ Diagnostic settings→  46‣ Properties→  46
SSD Out of range delay time→  46SSD Monitoring delay time→  46500 Process alert pressure→  46

Low alert value→  46

High alert value→  47501 Process alert scaled variable→  47

Low alert value→  47

High alert value→  47User temperature process alert→  48Low alert value→  48High alert value→  48‣ Configuration→  48‣ Configuration10→  48436 Diagnostic behavior→  48436 Event category→  49500 Diagnostic behavior→  49500 Diagnostic behavior→  49500 Diagnostic behavior→  49500 Diagnostic behavior→  49500 Event category→  49500 Event category→  49Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Overview of the operating menu

500 Event category→  49500 Event category→  49501 Diagnostic behavior→  50501 Diagnostic behavior→  50501 Diagnostic behavior→  50501 Diagnostic behavior→  50501 Diagnostic behavior→  50501 Diagnostic behavior→  50501 Event category→  50501 Event category→  50501 Event category→  50501 Event category→  50502 Diagnostic behavior→  50502 Diagnostic behavior→  50502 Diagnostic behavior→  50502 Diagnostic behavior→  50502 Diagnostic behavior→  50502 Diagnostic behavior→  50502 Event category→  51502 Event category→  51502 Event category→  51502 Event category→  51‣ ProcessEndress+Hauser→  51
Sensor pressure range behavior→  51

Sensor pressure range behavior→  51

11

Overview of the operating menu

Deltabar PMD55B PROFINET with Ethernet-APL→  51900 Event category→  52900 Event category→  52900 Event category→  52900 Event category→  52900 Diagnostic behavior→  52900 Diagnostic behavior→  52900 Diagnostic behavior→  52900 Diagnostic behavior→  52906 Diagnostic behavior→  52906 Diagnostic behavior→  52906 Diagnostic behavior→  52906 Diagnostic behavior→  52906 Event category→  53906 Event category→  53906 Event category→  53906 Event category→  53→  54

Application‣ Measured values

12841 Event category→  54Pressure→  54

Scaled variable→  54

Sensor temperature→  54

Terminal voltage 1→  54

Electronics temperature→  54Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Overview of the operating menu

‣ Measuring units→  55Pressure unit

→  55

Decimal places pressure→  55Temperature unit

→  55

Scaled variable unit

→  56

Free text→  57

Decimal places scaled variable→  57‣ PROFINET→  66‣ Configuration→  66
PROFINET device name→  66

PROFINET device name→  66

Parameter change acknowledge mode→  67

Acknowledge parameter change→  67

Descriptor→  67‣ Analog input→  67‣ Analog input 1 to 7
Process value→  67

Assign process variable→  68Damping→  68‣ Binary input→  68‣ Binary input 1 to 2
Controller input value‣ Binary output

Endress+Hauser→  67→  68→  68→  69Set point value→  69BO block output value→  6913

Overview of the operating menu

Deltabar PMD55B PROFINET with Ethernet-APL

Failure behavior→  69Failure behavior delay→  69Fixed value→  70‣ Information→  70Device ID→  70PA profile version→  70‣ Application relation
AR state→  70

MAC address IO controller→  71

MAC address backup IO controller→  71

IP address IO controller→  71

IP address backup IO controller→  71→  72

System‣ Device management→  72

Device tag→  72

Locking status→  72

Configuration counter→  73

Reset device→  73‣ User management14→  70→  74User role→  74

Password→  74Enter access code→  74

Status password entry→  74

New password→  75

Confirm new password→  75Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Overview of the operating menu

Status password entry→  74

Old password→  75

New password→  75

Confirm new password→  75

Status password entry→  74

Reset password→  75

Status password entry→  74

Old password→  75

Status password entry→  74‣ Connectivity→  76‣ Interfaces→  76
Display operation→  76Web server functionality→  76

Bluetooth activation→  76

Service (UART-CDI)→  76‣ Ethernet

Endress+Hauser→  77
MAC address→  77

IP address→  77Subnet mask

→  77

Default gateway→  77

Service IP active→  78Interface connection status→  78Interface speed→  78Duplex status→  78

Auto negotiation status→  7815

Overview of the operating menu

Deltabar PMD55B PROFINET with Ethernet-APL‣ Display

16

Received packet number→  79Sent packet number→  79Number of failed received packets→  79Number of failed sent packets→  79Signal to noise ratio→  79Number of failed received packets→  79Active TCP connections→  80Supported TCP connections→  80

TCP connection requests→  80

TCP connection timeouts→  80

Number of TCP connections closed→  80

Number of received TCP packets→  80

TCP sent packet number→  81

Number of TCP failed received packets→  81

Available UDP ports→  81

UDP received packet number→  81

UDP sent packet number→  81

Number of UDP failed received packets→  81→  82

Language→  82Format display→  82

Value 1 display→  83

Value 2 display→  83

Value 3 display→  84Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Value 4 display→  84Contrast display→  85‣ Date/time→  85Date/time→  85Time zone→  85

Enable NTP→  86

NTP server address→  87Clock synchronized→  87‣ Geolocation→  87
Location description→  87

Longitude→  87

Latitude→  88

Altitude→  88‣ Information

Endress+Hauser

Overview of the operating menu

→  88Device name→  88Manufacturer→  88

Serial number→  89Order code→  89Firmware version→  89Hardware version→  90Extended order code 1→  90Extended order code 2→  9017Overview of the operating menu

Deltabar PMD55B PROFINET with Ethernet-APL

Extended order code 3→  90

Checksum→  91‣ Software configuration→  91

Activate SW option→  91

Software option overview→  92‣ Firmware update→  92‣ Start update→  92
I have read the warning texts.

‣ Device information→  93

Device tag→  93

Device name→  94

Package version→  94‣ Select file→  94File check status‣ Performing verification
File validation status‣ Finish→  94→  95→  95→  95

I have read the warning texts.

18→  93→  95Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

3

Description of device parameters

Description of device parameters

In the following section, the parameters are listed according to the menu structure of theoperating tool.
The operating menu is dynamic and adapts the choice of parameters to the selected
options.

3.1

User navigation

The Guidance main menu contains functions which enable users to perform basic tasks
swiftly, e.g. commissioning.
These are primarily guided wizards and cross-subject special functions.
Navigation

3.1.1

 Guidance

Overview of the operating menu

User navigation• Commissioning (→  23)• Import/Export→  20
• Compare→  21

3.1.2

Commissioning

Run this wizard to put the device into operation. Enter the appropriate value in eachparameter or select the appropriate option.
If the wizard is canceled before all the necessary parameters have been configured,
any settings already made are saved. For this reason, the device may then be in an
undefined state!
In such situations, it is advisable to reset the device to the factory default settings.

Endress+Hauser

19Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL

The following parameters are configured in the Commissioning wizard:• Device identification (→  23)
This page summarizes the most important data characterizing the device. Some of theparameters can be edited, others are displayed for information only.
• Device tag (→  23)• Device name (→  23)• Serial number (→  23)• Extended order code 1 (→  24)• Extended order code 2 (→  24)• Extended order code 3 (→  24)• Locking status (→  25)• Time zone (→  26)• Date/time (→  27)• PROFINET device name (→  27)• IP address (→  27)• Descriptor (→  28)• MAC address (→  28)• Device ID (→  28)• Manufacturer ID (→  28)• Measurement adjustments (→  29)• Damping (→  29)• Assign scaled variable? (→  29)• Pressure unit (→  29)• Temperature unit (→  30)• Scaled variable unit (→  31)• Free text (→  31)• Temperature unit (→  30)• Temperature unit (→  30)• Zero adjustment (→  32)• Pressure (→  22)• Output settings (→  33)• Scaled variable transfer function (→  33)• Low flow cut off (→  33)• Lower Range Limit (→  34)• Upper Range Limit (→  34)• Minimum span (→  34)• Pressure (→  34)• Scaled variable (→  35)• Pressure value 1 (→  35)• Scaled variable value 1 (→  36)• Pressure value 2 (→  36)• Scaled variable value 2 (→  36)• Assign process variable (→  37)

3.1.3

Import / Export

Save / Load
• Save: The device settings can be saved in a .deh file.
• Load: The device settings saved in a .deh file can be written to the device.
Create documentation• Device documentation can be saved in PDF format under "Create documentation".
• This documentation contains the following general device information:• Information on device parameters• Information on linearization• Echo curves• Event list
• Diagnostic list

20

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

3.1.4

Description of device parameters

CompareCompare datasets
This function can be used to compare the following datasets:• Datasets in the .deh file format from the import/export function• Datasets with the configuration currently in the device

Endress+Hauser

21Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL

3.2"Device information" menu

Navigation

 Device info

Status signal
Navigation

Device info → Status signal

User interface• OK• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• --• Not categorized

Pressure
Navigation

Device info → Pressure

Navigation

Device info → Scaled variable

User interface

Signed floating-point number

Scaled variable

Do not show this message again
Navigation

Selection

Yes

22

Device info → Don't show again

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

3.3

Description of device parameters"Guidance" menu

Navigation

3.3.1

 Guidance"Commissioning" wizard

Navigation

 Guidance → Commissioning"Device identification" wizard
Navigation

 Guidance → Commissioning → Device ident.

Device tag
Navigation

Guidance → Commissioning → Device ident. → Device tag

Description

Enter a name for the measuring point to identify the measuring device in the plantUser entry

Character string comprising numbers, letters and special characters (32)

Device name
Navigation

Guidance → Commissioning → Device ident. → Device name

Description

Displays the name of the transmitter. It can also be found on the nameplate of thetransmitter.

User interface

Max. 32 characters such as letters or numbers.

Serial number
Navigation

Guidance → Commissioning → Device ident. → Serial number

Description

Displays the serial number of the measuring device.
The number can be found on the nameplate of the sensor and transmitter.

User interface

Endress+Hauser

Max. 11-digit character string comprising letters and numbers.

23

Description of device parameters

Additional information

Deltabar PMD55B PROFINET with Ethernet-APL

Description
Uses of the serial number• To identify the measuring device quickly, e.g. when contacting Endress+Hauser.
• To obtain specific information on the measuring device using the Device Viewer:
www.endress.com/deviceviewer

Extended order code 1Navigation

Guidance → Commissioning → Device ident. → Ext. order cd. 1

Description

The extended order code is an alphanumeric code containing all information to identify
the device and its options.

User interface

Character string

Factory setting–

Additional information

Description
The extended order code indicates the version of all the features of the product structure
for the measuring device and thus uniquely identifies the measuring device.

Extended order code 2Navigation

Guidance → Commissioning → Device ident. → Ext. order cd. 2

Description

The extended order code is an alphanumeric code containing all information to identify
the device and its options.
The extended order code can also be found on the nameplate of the sensor andtransmitter in the "Ext. ord. cd." field.

User interface

Character string

Factory setting–

Extended order code 3

Navigation

Guidance → Commissioning → Device ident. → Ext. order cd. 3

Description

The extended order code is an alphanumeric code containing all information to identify
the device and its options.
The extended order code can also be found on the nameplate of the sensor andtransmitter in the "Ext. ord. cd." field.

User interface

24

Character string

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Factory setting

Description of device parameters–"Device identification" wizard
Navigation

 Guidance → Commissioning → Device ident.

Locking status
Navigation

Guidance → Commissioning → Device ident. → Locking status

Description

Displays the active write protection.

User interface• Hardware locked• Temporarily locked

Additional information

User interface
If two or more types of write protection are active, the write protection with the highest
priority is shown on the local display. In the operating tool all active types of write
protection are displayed.
Detailed information on access authorization is provided in the "User roles andassociated access authorization" and "Operating concept" sections of the Operations
Instructions for the device.
Selection
Function scope of the "Locking status" parameter

Endress+Hauser

Options

Description

None

The access status displayed in the Access status display parameter applies. Only
appears on local display.

Hardware locked

The DIP switch for hardware locking is activated on the main electronics module.
This prevents write access to the parameters (e.g. via the local display or operatingtool).

Temporarily locked

Write access to the parameters is temporarily locked due to device-internal
processing (e.g. data upload/download, reset). Once the internal processing has
been completed, the parameters can be changed once again.

25

Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL"Device identification" wizard
Navigation

 Guidance → Commissioning → Device ident.

Time zoneNavigationDescription

Select the time zone. Every time the time zone is changed, a logbook entry is created.

Selection

26

Guidance → Commissioning → Device ident. → Time zone

Other units• UTC-12:00• UTC-11:00• UTC-10:00• UTC-09:30• UTC-09:00• UTC-08:00• UTC-07:00• UTC-06:00• UTC-05:00• UTC-04:00• UTC-03:30• UTC-03:00• UTC-02:30• UTC-02:00• UTC-01:00• UTC 00:00• UTC+01:00• UTC+02:00• UTC+03:00• UTC+03:30• UTC+04:00• UTC+04:30• UTC+05:00• UTC+05:30• UTC+05:45• UTC+06:00• UTC+06:30• UTC+07:00• UTC+08:00• UTC+08:45• UTC+09:00• UTC+09:30• UTC+10:00• UTC+10:30• UTC+11:00• UTC+12:00• UTC+12:45• UTC+13:00• UTC+13:45• UTC+14:00

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters

Date/time
Navigation

Guidance → Commissioning → Device ident. → Date/time

Description

Displays the date and time entered.

User interface

Character string comprising numbers, letters and special characters"Device identification" wizard
Navigation

 Guidance → Commissioning → Device ident.

PROFINET device name
Navigation

Guidance → Commissioning → Device ident. → PROFINET DevName

Description

Up to 240 characters are allowed.
The following syntax must be used:- 1 or more identifiers, separated with [.]- Identifier length is 1 to 63 characters- Identifier consists of [a-z 0-9] only lowercase letters and numbers allowed.

User entry

Character string comprising numbers, letters and special characters (240)

PROFINET device name
Navigation

Guidance → Commissioning → Device ident. → PROFINET DevName

Description

Shows the short form of the PROFINET device name for the measuring point

User interface

Character string comprising numbers, letters and special characters

IP addressNavigationDescription

Enter the IP address of the measuring device

User entry

Character string comprising numbers, letters and special characters (15)Endress+Hauser

Guidance → Commissioning → Device ident. → IP address27

Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL

Descriptor
Navigation

Guidance → Commissioning → Device ident. → Descriptor

Description

Enter a description for the measuring point

User entry

Character string comprising numbers, letters and special characters (54)

MAC address
Navigation

Guidance → Commissioning → Device ident. → MAC Address

Description

Shows the MAC address of the measuring device

User interface

Character string comprising numbers, letters and special characters

Device ID
Navigation

Guidance → Commissioning → Device ident. → Device ID

User interface

0 to 65 535

Manufacturer ID
Navigation

User interface

0 to 65 535

28

Guidance → Commissioning → Device ident. → Manufacturer ID

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters"Measurement adjustments" wizard
Navigation

 Guidance → Commissioning → Meas. adjust.

DampingNavigation

Guidance → Commissioning → Meas. adjust. → Damping

Description

The damping is effective before the measured value is further processed, i.e., before thefollowing processes:- Scaling- Limit value monitoring- Forwarding to display- Forwarding to Analog Input BlockNote:
The Analog Input Block has its own “Damping” parameter. In the measurement chain, only
one of the two attenuation parameters shall have a value other than 0.
Otherwise, the signal will be attenuated several times.

User entry

0 to 999.0 s

Assign scaled variable?



Navigation

Selection

• No
• Yes

Guidance → Commissioning → Meas. adjust. → Scaled variable?

"Measurement adjustments" wizard
Navigation

 Guidance → Commissioning → Meas. adjust.

Pressure unitNavigationDescription

Use this function to select the unit for the pipe pressure.

Endress+Hauser

Guidance → Commissioning → Meas. adjust. → Pressure unit

29Description of device parameters

Selection

Deltabar PMD55B PROFINET with Ethernet-APL

SI units
• MPa
• kPa
• Pa
• bar
• mbar a• torr• atm• kgf/cm²• gf/cm²

US unitspsi

Other units• inH2O• inH2O (4°C)• mmH2O• mmH2O (4°C)• mH2O• mH2O (4°C)• ftH2O• inHg• mmHgTemperature unitNavigation

Guidance → Commissioning → Meas. adjust. → Temperature unit

Description

Use this function to select the unit for the temperature.

Selection

SI units• °C•KFactory setting

Country-specific:
• °C
• °F

Additional information

Selection

US units°F"Measurement adjustments" wizard
Navigation

 Guidance → Commissioning → Meas. adjust.

Pressure unitNavigationDescription

Use this function to select the unit for the pipe pressure.

Selection

SI units
• MPa
• kPa
• Pa
• bar
• mbar a• torr• atm• kgf/cm²• gf/cm²

30

Guidance → Commissioning → Meas. adjust. → Pressure unit

US unitspsi

Other units• inH2O• inH2O (4°C)• mmH2O• mmH2O (4°C)• mH2O• mH2O (4°C)• ftH2O• inHg• mmHgEndress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters

Scaled variable unitNavigation

Guidance → Commissioning → Meas. adjust. → SV unit

Description

Use 'Free text', first selection, if the desired unit is not available in the selection list. It is
possible to define a customer specific unit with another parameter.

Selection

SI units
•%
• mm
• cm
•m
•l
• hl
• m³
•g
• kg
•t
• g/s
• kg/s• kg/min• kg/h• t/min• t/h• t/d• m³/s• m³/min• m³/h• m³/d• l/s• l/min• l/h• Nm³/h• Nl/h• Sm³/s• Sm³/min• Sm³/h• Sm³/d• Nm³/s• g/cm³• kg/m³• Nm³/min• Nm³/dUS units
• ft
• in
• ft³
• gal (us)• bbl (us;oil)
• oz
• lb
• STon• lb/s• lb/min• lb/h• STon/min• STon/h• STon/d• ft³/s• ft³/min• ft³/h• ft³/d• gal/s (us)• gal/min (us)• gal/h (us)• gal/d (us)• bbl/s (us;oil)• bbl/min (us;oil)• bbl/h (us;oil)• bbl/d (us;oil)• Sft³/min• Sft³/h• Sft³/d

Imperial units• gal (imp)• gal/s (imp)• gal/min (imp)• gal/h (imp)

Custom-specific units
Free text

Free text

Navigation

User entry

Character string comprising numbers, letters and special characters (32)Endress+Hauser

Guidance → Commissioning → Meas. adjust. → Free text

31

Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL

Temperature unitNavigation

Guidance → Commissioning → Meas. adjust. → Temperature unit

Description

Use this function to select the unit for the temperature.

Selection

SI units• °C•KFactory setting

Country-specific:
• °C
• °F

Additional information

Selection

US units°F"Measurement adjustments" wizard
Navigation

 Guidance → Commissioning → Meas. adjust.

Zero adjustment

NavigationDescription

Due to the mounting position of the measuring instrument, a pressure shift may occur.
The pressure shift can be corrected with the zero adjustment.

Selection• No• Confirm

32

Guidance → Commissioning → Meas. adjust. → Zero adjustment

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters

Pressure
Navigation

Guidance → Commissioning → Meas. adjust. → Pressure"Output settings" wizard
Navigation

 Guidance → Commissioning → Output settings

Scaled variable transfer function



Navigation

Guidance → Commissioning → Output settings → Scal. v. trans.

Description'Linear'
The linear pressure signal is used for the output signal. The flow must be calculated in theevaluation unit. Deviating from the bar graph (output signal), the digital value on thedisplay shows continues to be the eradicated value.
'Square root'
The root flow signal is used for the output signal. The 'Flow (square root)'
output signal is indicated on the on-site display with a root symbol.
'Table'The output is defined according to the scaled variable / pressure table entered.

Selection• Linear• Square root *• TableLow flow cut offNavigationDescription

When activated, this function suppresses small flows which can lead to large fluctuationsin the measured value.

User entry

0.0 to 50.0 %*Guidance → Commissioning → Output settings → Low flow cut off

Visibility depends on order options or device settings

Endress+Hauser

33

Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL"Output settings" wizard
Navigation

 Guidance → Commissioning → Output settings

Lower Range Limit
Navigation

Guidance → Commissioning → Output settings → LRL

Description

Indicates the lower measuring limit of the sensor.

User interface

Signed floating-point number

Upper Range Limit
Navigation

Guidance → Commissioning → Output settings → URL

Description

Indicates the upper measuring limit of the sensor.

User interface

Signed floating-point number

Minimum span
Navigation

Guidance → Commissioning → Output settings → Minimum span

Description

Specifies the smallest possible measuring span of the sensor.

User interface

Signed floating-point number"Output settings" wizard
Navigation

 Guidance → Commissioning → Output settings

Pressure

Navigation

User entry

Signed floating-point number

34

Guidance → Commissioning → Output settings → Pressure

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters

Scaled variable

Navigation

Guidance → Commissioning → Output settings → Scaled variable

User entry

Signed floating-point number"Output settings" wizard
Navigation

 Guidance → Commissioning → Output settings

Scaled variable transfer function



Navigation

Guidance → Commissioning → Output settings → Scal. v. trans.

Description'Linear'
The linear pressure signal is used for the output signal. The flow must be calculated in theevaluation unit. Deviating from the bar graph (output signal), the digital value on thedisplay shows continues to be the eradicated value.
'Square root'
The root flow signal is used for the output signal. The 'Flow (square root)'
output signal is indicated on the on-site display with a root symbol.
'Table'The output is defined according to the scaled variable / pressure table entered.

Selection• Linear• Square root *• TablePressure value 1NavigationDescription

Enter pressure for the first scaling point. 'Scaled variable value 1' will be allocated to thispressure.

User entry

Signed floating-point number*

Guidance → Commissioning → Output settings → P. value 1

Visibility depends on order options or device settings

Endress+Hauser

35Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL

Scaled variable value 1
Navigation

Guidance → Commissioning → Output settings → Sc. var.value 1

Description

Enter value for the first scaling point. This value is allocated to 'Pressure value 1'.

User interface

Signed floating-point number

Pressure value 2Navigation

Guidance → Commissioning → Output settings → P. value 2

Description

Enter pressure for the second scaling point. 'Scaled variable value 2' will be allocated to thispressure.

User entry

Signed floating-point number

Scaled variable value 2Navigation

Guidance → Commissioning → Output settings → Sc. var.value 2

Description

Enter value for the second scaling point. This value is allocated to 'Pressure value 2'.

User entry

Signed floating-point number

Lower Range Limit
Navigation

Guidance → Commissioning → Output settings → LRL

Description

Indicates the lower measuring limit of the sensor.

User interface

Signed floating-point number

Upper Range Limit
NavigationDescription

Indicates the upper measuring limit of the sensor.

User interface

Signed floating-point number

36

Guidance → Commissioning → Output settings → URL

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters

Minimum span
Navigation

Guidance → Commissioning → Output settings → Minimum span

Description

Specifies the smallest possible measuring span of the sensor.

User interface

Signed floating-point number"Output settings" wizard
Navigation

 Guidance → Commissioning → Output settings

Assign process variable
Navigation

Guidance → Commissioning → Output settings → Assign variableGuidance → Commissioning → Output settings → Assign variableGuidance → Commissioning → Output settings → Assign variableGuidance → Commissioning → Output settings → Assign variableGuidance → Commissioning → Output settings → Assign variableGuidance → Commissioning → Output settings → Assign variableGuidance → Commissioning → Output settings → Assign variable

Description

Select a process variable

User interface• Pressure *• Scaled variable *• Sensor temperature *• Sensor pressure *• Electronics temperature *• Median of pressure signal *• Noise of pressure signal *Additional information

User interface"Sensor pressure" option
Sensor Pressure is the raw signal from sensor before damping and position adjustment.

*

Visibility depends on order options or device settings

Endress+Hauser

37Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL

3.4"Diagnostics" menu

Navigation

3.4.1



Diagnostics"Active diagnostics" submenu

Navigation

Diagnostics → Active diagnos.

Active diagnostics
Navigation

 Diagnostics → Active diagnos. → Active diagnos.

Prerequisite

A diagnostic event has occurred.

Description

Displays the current diagnostic message. If two or more messages occur simultaneously,
the message with the highest priority is shown on the display.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

Additional information

User interface
Additional pending diagnostic messages can be viewed in the Diagnostic list
submenu.
ExampleFor the display format:
F271 Main electronic failure

Timestamp
Navigation

 Diagnostics → Active diagnos. → Timestamp

Description

Displays the operating time when the current diagnostic message occurred.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

User interface
The diagnostic message can be viewed via the Actual diagnostics parameter(→  38).
ExampleFor the display format:
24d12h13m00s

38

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters

Previous diagnostics
Navigation

 Diagnostics → Active diagnos. → Prev.diagnostics

Prerequisite

Two diagnostic events have already occurred.

Description

Displays the diagnostic message that occurred before the current message.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

Additional information

User interface
Via the local display: the time stamp and corrective measures referring to the cause of
the diagnostic message can be accessed via the  key.
ExampleFor the display format:
F271 Main electronic failure

Timestamp
Navigation

 Diagnostics → Active diagnos. → Timestamp

Description

Displays the operating time when the last diagnostic message before the current message
occurred.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

User interface
The diagnostic message can be viewed via the Previous diagnostics parameter(→  39).
ExampleFor the display format:
24d12h13m00s

Operating time from restart
Navigation

 Diagnostics → Active diagnos. → Time fr. restart

Description

Indicates how long the device has been in operation since the last time the device wasrestarted.

User interface

Days (d), hours (h), minutes (m), seconds (s)Endress+Hauser

39Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL

Operating time
Navigation

 Diagnostics → Active diagnos. → Operating time

Description

Indicates how long the device has been in operation.

Additional information

Maximum time: 9 999 d (≈ 27 years)

3.4.2"Event logbook" submenu

Navigation

Diagnostics → Event logbook

Filter options
Navigation

Diagnostics → Event logbook → Filter options

Description

Use this function to select the category whose event messages are displayed in the event
list of the operating tool.

Selection• All• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• Information (I)• Not categorized

Additional information

Description
The status signals are categorized in accordance with VDI/VDE 2650 and NAMUR
Recommendation NE 107:
• F = Failure• C = Function Check• S = Out of Specification• M = Maintenance Required

40

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

3.4.3

Description of device parameters"Minimum/maximum values" submenu

Navigation

Diagnostics → Min/max val.

Pressure min
Navigation

 Diagnostics → Min/max val. → Pressure min

Description

Minimum or maximum value measured by device.

User interface

Signed floating-point number

Counter limit underruns sensor Pmin
Navigation

 Diagnostics → Min/max val. → Counter P < Pmin

Description

Counts how many times the value underruns the sensor specific minimum values.
Sensor specific minimum values are shown in Application/Sensor menu.

User interface

0 to 65 535Counter underruns of user limit Pmin
Navigation

 Diagnostics → Min/max val. → Counter < P user

Description

Counts how many times the value underruns the minimum values defined by the user.
User defined minimum values are shown in Diagnostic/Diagnostic settings/Properties
menu.

User interface

0 to 65 535

Minimum sensor temperature
Navigation

 Diagnostics → Min/max val. → Min. sensor temp

Description

Minimum or maximum value measured by device.
Users cannot reset this value.

User interface–273.15 to 9 726.85 °C

Endress+Hauser

41Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL

Counter limit underruns sensor Tmin
Navigation

 Diagnostics → Min/max val. → Counter T < Tmin

Description

Counts how many times the value underruns/overruns the sensor specific minimum/
maximum values.
Sensor specific minimum/maximum values are shown in Application/Sensor menu.

User interface

0 to 65 535Counter underruns of user limit Tmin
Navigation

 Diagnostics → Min/max val. → Counter < T user

User interface

0 to 65 535

Minimum terminal voltage
Navigation

 Diagnostics → Min/max val. → Min.term.volt.

Description

Minimum or maximum measured terminal (supply) voltage.

User interface

0.0 to 50.0 V

Minimum electronics temperature
Navigation

 Diagnostics → Min/max val. → Min.electr.temp.

Description

Minimum or maximum measured main electronics temperature.

User interface

Signed floating-point number

Reset user defined counters P and T
Navigation

 Diagnostics → Min/max val. → Reset count. P T

Selection• Cancel• Confirm42Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters

Pressure max
Navigation

 Diagnostics → Min/max val. → Pressure max

Description

Minimum or maximum value measured by device.

User interface

Signed floating-point number

Counter limit overruns sensor Pmax
Navigation

 Diagnostics → Min/max val. → Counter P > Pmax

Description

Counts how many times the value overruns the sensor specific maximum values.
Sensor specific maximum values are shown in Application/Sensor menu.

User interface

0 to 65 535Counter overruns of user limit Pmax
Navigation

 Diagnostics → Min/max val. → Counter > P user

Description

Counts how many times the value overruns the maximum values defined by the user.
User defined maximum values are shown in Diagnostic/Diagnostic settings/Properties
menu.

User interface

0 to 65 535

Maximum sensor temperature
Navigation

 Diagnostics → Min/max val. → Max. sensor temp

Description

Minimum or maximum value measured by device.
Users cannot reset this value.

User interface–273.15 to 9 726.85 °C

Endress+Hauser

43Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL

Counter limit overruns sensor Tmax
Navigation

 Diagnostics → Min/max val. → Counter T > Tmax

Description

Counts how many times the value underruns/overruns the sensor specific minimum/
maximum values.
Sensor specific minimum/maximum values are shown in Application/Sensor menu.

User interface

0 to 65 535Counter overruns of user limit Tmax
Navigation

 Diagnostics → Min/max val. → Counter > T user

User interface

0 to 65 535

Maximum terminal voltage
Navigation

 Diagnostics → Min/max val. → Max.term.voltage

Description

Minimum or maximum measured terminal (supply) voltage.

User interface

0.0 to 50.0 V

Maximum electronics temperature
Navigation

 Diagnostics → Min/max val. → Max.electr.temp.

Description

Minimum or maximum measured main electronics temperature.

User interface

Signed floating-point number

44

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

3.4.4
Navigation

Description of device parameters"Simulation" submenu
Diagnostics → Simulation

Simulation



Navigation

 Diagnostics → Simulation → Simulation

Description

Simulates one or more process variables and/or events.
Warning:
Output will reflect the simulated value or event.

Selection• Off• Pressure• Diagnostic event simulation

Diagnostic event simulation



Navigation

 Diagnostics → Simulation → Diag. event sim.

Description

Use this function to select a diagnostic event for the simulation process that is activated.

Selection• Off• Diagnostic event picklist (depends on the category selected)

Additional information

Description
For the simulation, you can choose from the diagnostic events of the category selectedin the Diagnostic event category parameter.

Value pressure simulation



Navigation

 Diagnostics → Simulation → Value pressure

User entry

Signed floating-point number

Endress+Hauser

45Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL

3.4.5"Diagnostic settings" submenu

Navigation

Diagnostics → Diag. settings"Properties" submenu
Navigation

Diagnostics → Diag. settings → Properties

SSD Out of range delay timeNavigation

Diagnostics → Diag. settings → Properties → SSD Delay time

User entry

0 to 604 800 s

SSD Monitoring delay timeNavigation

Diagnostics → Diag. settings → Properties → SSD Verz. Zeit

User entry

0 to 86 400 s

500 Process alert pressureNavigation

Diagnostics → Diag. settings → Properties → 500 Pressure

Description

Define whether user-defined pressure limits should be set.
If 'No' is selected, no analysis will take place and no event message will be generated.

Selection

• Off
• On

Low alert value

NavigationDescription

Set area.
If this limit value is exceeded or undercut, an event is generated. There is no hysteresis.

User entry

Signed floating-point number

46

Diagnostics → Diag. settings → Properties → Low alert value

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters

High alert value

Navigation

Diagnostics → Diag. settings → Properties → High alert value

Description

Set area.
If this limit value is exceeded or undercut, an event is generated. There is no hysteresis.

User entry

Signed floating-point number

501 Process alert scaled variable

Navigation

Diagnostics → Diag. settings → Properties → 501 Scaled var.

Description

Define whether user-defined limits should be set.
If 'No' is selected, no analysis will take place and no event message will be generated.

Selection

• Off
• On

Low alert value

Navigation

Diagnostics → Diag. settings → Properties → Low alert value

Description

Set area.
If this limit value is exceeded or undercut, an event is generated. There is no hysteresis.

User entry

Signed floating-point number

High alert value

NavigationDescription

Set area.
If this limit value is exceeded or undercut, an event is generated. There is no hysteresis.

User entry

Signed floating-point number

Endress+Hauser

Diagnostics → Diag. settings → Properties → High alert value

47

Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL

User temperature process alert

Navigation

Diagnostics → Diag. settings → Properties → UserTemp alert

Description

Define whether the user-defined sensor temperature limits should be set. If 'No' no analysis
and therefore no event message will take place.

Selection

• Off
• On

Low alert value

Navigation

Diagnostics → Diag. settings → Properties → Low alert value

Description

Set area.
If this limit value is exceeded or undercut, an event is generated. There is no hysteresis.

User entry–50 to 150 °C

High alert value

Navigation

Diagnostics → Diag. settings → Properties → High alert value

Description

Set area.
If this limit value is exceeded or undercut, an event is generated. There is no hysteresis.

User entry–50 to 150 °C"Configuration" submenu
Navigation



436 Diagnostic behavior

Navigation

Selection• Warning• Logbook entry only

48

Diagnostics → Diag. settings → Configuration → Configuration

Diagnostics → Diag. settings → Configuration → Configuration → 436 Diag. behav.

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters

436 Event category
Navigation

Diagnostics → Diag. settings → Configuration → Configuration → 436 Event
category

User interface• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• Not categorized

500 Diagnostic behavior

Navigation

 Diagnostics → Diag. settings → Configuration → Configuration → 500 Diag. behav.

Description

Select event behavior'Logbook entry only':
no digital or analog transmission of the message'Warning': Current output unchanged. Message is output digitally (default).
'Alarm': Current output assumes the set alarm current.
Regardless of the setting, the message appears on the display. If the permissible conditions
are reached again, the warning is no longer available in the instrument.

Selection• Off• Alarm• Warning• Logbook entry only

500 Event category
Navigation

 Diagnostics → Diag. settings → Configuration → Configuration → 500Event category

User interface• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• Not categorized

Endress+Hauser

49

Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL

501 Diagnostic behavior

Navigation

 Diagnostics → Diag. settings → Configuration → Configuration → 501 Diag. behav.

Description

Select event behavior'Logbook entry only':
no digital or analog transmission of the message'Warning': Current output unchanged. Message is output digitally (default).
'Alarm': Current output assumes the set alarm current.
Regardless of the setting, the message appears on the display. If the permissible conditions
are reached again, the warning is no longer available in the instrument.

Selection• Off• Alarm• Warning• Logbook entry only

501 Event category
Navigation

 Diagnostics → Diag. settings → Configuration → Configuration → 501Event category

User interface• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• Not categorized

502 Diagnostic behavior

Navigation

 Diagnostics → Diag. settings → Configuration → Configuration → 502 Diag. behav.

Description

Select event behavior'Logbook entry only':
no digital or analog transmission of the message'Warning': Current output unchanged. Message is output digitally (default).
'Alarm': Current output assumes the set alarm current.
Regardless of the setting, the message appears on the display. If the permissible conditions
are reached again, the warning is no longer available in the instrument.

Selection

50

• Off
• Alarm• Warning• Logbook entry only

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters

502 Event category
Navigation

 Diagnostics → Diag. settings → Configuration → Configuration → 502Event category

User interface• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• Not categorized"Process" submenu
Navigation

Diagnostics → Diag. settings → Configuration → Process

Sensor pressure range behavior

Navigation

 Diagnostics → Diag. settings → Configuration → Process → P-range behavior

Description

Alarm and warning messages appear on the display. If the permissible conditions arereached again, the warning message disappears.
Select event behavior:'Alarm':The signal outputs assume the specified alarm condition. A diagnostic message is
generated.
'Warning':
The signal outputs assume the specified warning condition. A diagnostic message is
generated.
'Logbook entry only':
No digital or analog forwarding of the message. A diagnostic message is written in thelogbook.

Selection• Alarm• Warning• Logbook entry only

841 Event category
Navigation

User interface• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• Not categorized

Endress+Hauser

Diagnostics → Diag. settings → Configuration → Process → 841 Event category

51

Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL

900 Event category
Navigation

 Diagnostics → Diag. settings → Configuration → Process → 900Event category

User interface• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• Not categorized

900 Diagnostic behavior

Navigation

 Diagnostics → Diag. settings → Configuration → Process → 900 Diag. behav.

Description

Select event behavior'Logbook entry only':No forwarding of the message via the fieldbus.
'Warning': Warning message is transmitted via the fieldbus (default setting).
Regardless of the setting, the message appears on the display. If the permissible conditions
are reached again, the warning is no longer available in the instrument.

Selection• Warning• Logbook entry only

906 Diagnostic behavior

Navigation

 Diagnostics → Diag. settings → Configuration → Process → 906 Diag. behav.

Description

Select event behavior'Logbook entry only':No forwarding of the message via the fieldbus.
'Warning': Warning message is transmitted via the fieldbus (default setting).
Regardless of the setting, the message appears on the display. If the permissible conditions
are reached again, the warning is no longer available in the instrument.

Selection

52

• Off
• Warning• Logbook entry only

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters

906 Event category
Navigation

 Diagnostics → Diag. settings → Configuration → Process → 906Event category

User interface• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• Not categorized

Endress+Hauser

53Description of device parameters

3.5

Deltabar PMD55B PROFINET with Ethernet-APL"Application" menu

Navigation

3.5.1



Application"Measured values" submenu

Navigation

Application → Measured values

Pressure
Navigation

 Application → Measured values → Pressure

Scaled variable
Navigation

 Application → Measured values → Scaled variable

User interface

Signed floating-point number

Sensor temperature
Navigation

 Application → Measured values → Sensor temp.

User interface–273.15 to 9 726.85 °C

Terminal voltage 1
Navigation

 Application → Measured values → Terminal volt. 1

Description

Shows the current terminal voltage that is applied at the output

User interface

0.0 to 50.0 V

Electronics temperature
Navigation

 Application → Measured values → Electronics temp

Description

Displays the current temperature of the main electronics.

54

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

User interface

Description of device parameters

Signed floating-point number

3.5.2"Measuring units" submenu

Navigation

Application → Measuring units

Pressure unitNavigation

 Application → Measuring units → Pressure unit

Description

Use this function to select the unit for the pipe pressure.

Selection

SI units
• MPa
• kPa
• Pa
• bar
• mbar a• torr• atm• kgf/cm²• gf/cm²

US unitspsi

Other units• inH2O• inH2O (4°C)• mmH2O• mmH2O (4°C)• mH2O• mH2O (4°C)• ftH2O• inHg• mmHg

Decimal places pressureNavigation

 Application → Measuring units → Decimal pressure

Description

This selection does not affect the measurement and calculation accuracy of the device.

Selection• Automatic•x• x.x• x.xx• x.xxx• x.xxxx

Temperature unitNavigation

 Application → Measuring units → Temperature unit

Description

Use this function to select the unit for the temperature.

Endress+Hauser

55Description of device parameters

Selection

SI units• °C•KFactory setting

Country-specific:
• °C
• °F

Additional information

Selection

Deltabar PMD55B PROFINET with Ethernet-APL

US units°F

Scaled variable unitNavigation

 Application → Measuring units → SV unit

Description

Use 'Free text', first selection, if the desired unit is not available in the selection list. It is
possible to define a customer specific unit with another parameter.

56

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Selection

SI units
•%
• mm
• cm
•m
•l
• hl
• m³
•g
• kg
•t
• g/s
• kg/s• kg/min• kg/h• t/min• t/h• t/d• m³/s• m³/min• m³/h• m³/d• l/s• l/min• l/h• Nm³/h• Nl/h• Sm³/s• Sm³/min• Sm³/h• Sm³/d• Nm³/s• g/cm³• kg/m³• Nm³/min• Nm³/dDescription of device parameters

US units
• ft
• in
• ft³
• gal (us)• bbl (us;oil)
• oz
• lb
• STon• lb/s• lb/min• lb/h• STon/min• STon/h• STon/d• ft³/s• ft³/min• ft³/h• ft³/d• gal/s (us)• gal/min (us)• gal/h (us)• gal/d (us)• bbl/s (us;oil)• bbl/min (us;oil)• bbl/h (us;oil)• bbl/d (us;oil)• Sft³/min• Sft³/h• Sft³/d

Imperial units• gal (imp)• gal/s (imp)• gal/min (imp)• gal/h (imp)

Custom-specific units
Free text

Free text

Navigation

 Application → Measuring units → Free text

User entry

Character string comprising numbers, letters and special characters (32)

Decimal places scaled variable
Navigation

 Application → Measuring units → Decimal scaled

Description

This selection does not affect the measurement and calculation accuracy of the device.

Endress+Hauser57Description of device parameters

Selection

Deltabar PMD55B PROFINET with Ethernet-APL

•x
• x.x
• x.xx• x.xxx• x.xxxx

3.5.3
Navigation"Sensor" submenu
Application → Sensor → Sensor cal.

Zero adjustment

Navigation

 Application → Sensor → Sensor cal. → Zero adjustment

Description

Due to the mounting position of the measuring instrument, a pressure shift may occur.
The pressure shift can be corrected with the zero adjustment.

Selection• No• Confirm

Calibration offset

Navigation

 Application → Sensor → Sensor cal. → Calibr offset

Prerequisite

Absolute pressure sensor

User entry

Signed floating-point number

Zero adjustment offset

Navigation

 Application → Sensor → Sensor cal. → Zero adj. offset

User entry

Signed floating-point number

Sensor Trim Reset

Navigation

 Application → Sensor → Sensor cal. → Sen. Trim Reset

Selection• No• Confirm

58

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters

Lower sensor trim

Navigation

 Application → Sensor → Sensor cal. → LowerSensor trim

Description

These two parameters allow a recalibration of the sensor, i.e., if you want to fit the sensor
to the measuring range. The highest accuracy is obtained when the value for the 'Lower
sensor trim' is as close as possible to 'LRV' (lower range value) and the value for 'Upper
sensor trim' as close as possible to 'URV' (upper range value).
There must be a known reference pressure when setting a new lower or upper sensor
characteristic curve value. The more accurate the reference pressure is during
recalibration, the higher the accuracy of the pressure transmitter later. A new value isassigned to the applied pressure using 'Lower sensor trim' and 'Upper sensor trim'
parameters.
Proceed as follows:- Apply reference pressure for lower range value ('LRV')- Enter the measured reference pressure at 'Lower sensor trim' and confirm- Apply reference pressure for upper range value ('URV')- Enter the measured reference pressure at 'Upper sensor trim' and confirm- The sensor is now calibrated

User entry

Signed floating-point number

Upper sensor trim

Navigation

 Application → Sensor → Sensor cal. → UpperSensor trim

User entry

Signed floating-point number

Lower range value

Navigation

 Application → Sensor → Sensor cal. → Lower range val.

Description

The calibrated span corresponds to the span between the LRV and URV.
Factory setting: 0 to URL.
Other calibrated spans can be ordered as customized span.

User entry

Signed floating-point number

Upper range value

Navigation

 Application → Sensor → Sensor cal. → Upper range val.

Description

The calibrated span corresponds to the span between the LRV and URV.
Factory setting: 0 to URL.
Other calibrated spans can be ordered as customized span.

Endress+Hauser

59Description of device parameters

User entry

Deltabar PMD55B PROFINET with Ethernet-APL

Signed floating-point number"Sensor configuration" submenu
Navigation

Application → Sensor → Sensor conf.

DampingNavigation

 Application → Sensor → Sensor conf. → Damping

Description

The damping is effective before the measured value is further processed, i.e., before thefollowing processes:- Scaling- Limit value monitoring- Forwarding to display- Forwarding to Analog Input BlockNote:
The Analog Input Block has its own “Damping” parameter. In the measurement chain, only
one of the two attenuation parameters shall have a value other than 0.
Otherwise, the signal will be attenuated several times.

User entry

0 to 999.0 s

HP/LP swap

Navigation

 Application → Sensor → Sensor conf. → HP/LP swap

Description

With this parameter the high and low pressure side of the differential pressure transmittercan be interchanged.

Selection

• No
• Yes

60

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters"Sensor limits" submenu
Navigation

Application → Sensor → Sensor limits

Lower Range Limit
Navigation

Application → Sensor → Sensor limits → LRL

Description

Indicates the lower measuring limit of the sensor.

User interface

Signed floating-point number

Upper Range Limit
Navigation

Application → Sensor → Sensor limits → URL

Description

Indicates the upper measuring limit of the sensor.

User interface

Signed floating-point number

Minimum span
Navigation

Application → Sensor → Sensor limits → Minimum span

Description

Specifies the smallest possible measuring span of the sensor.

User interface

Signed floating-point number

Sensor temperature lower range limit
Navigation

User interface–273.15 to 9 726.85 °C

Endress+Hauser

Application → Sensor → Sensor limits → Sens.temp.lo.lim

61

Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL

Sensor temperature upper range limit
Navigation

Application → Sensor → Sensor limits → Sens.temp.up.lim

User interface–273.15 to 9 726.85 °C"Scaled variable" submenu
Navigation

Application → Sensor → Scaled variable

Scaled variable unitNavigation

 Application → Sensor → Scaled variable → SV unit

Description

Use 'Free text', first selection, if the desired unit is not available in the selection list. It is
possible to define a customer specific unit with another parameter.

62

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Selection

SI units
•%
• mm
• cm
•m
•l
• hl
• m³
•g
• kg
•t
• g/s
• kg/s• kg/min• kg/h• t/min• t/h• t/d• m³/s• m³/min• m³/h• m³/d• l/s• l/min• l/h• Nm³/h• Nl/h• Sm³/s• Sm³/min• Sm³/h• Sm³/d• Nm³/s• g/cm³• kg/m³• Nm³/min• Nm³/dDescription of device parameters

US units
• ft
• in
• ft³
• gal (us)• bbl (us;oil)
• oz
• lb
• STon• lb/s• lb/min• lb/h• STon/min• STon/h• STon/d• ft³/s• ft³/min• ft³/h• ft³/d• gal/s (us)• gal/min (us)• gal/h (us)• gal/d (us)• bbl/s (us;oil)• bbl/min (us;oil)• bbl/h (us;oil)• bbl/d (us;oil)• Sft³/min• Sft³/h• Sft³/d

Imperial units• gal (imp)• gal/s (imp)• gal/min (imp)• gal/h (imp)

Custom-specific units
Free text

Free text

Navigation

 Application → Sensor → Scaled variable → Free text

User entry

Character string comprising numbers, letters and special characters (32)Endress+Hauser

63Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL

Pressure
Navigation

 Application → Sensor → Scaled variable → Pressure

Scaled variable transfer function



Navigation

 Application → Sensor → Scaled variable → Scal. v. trans.

Description'Linear'
The linear pressure signal is used for the output signal. The flow must be calculated in theevaluation unit. Deviating from the bar graph (output signal), the digital value on thedisplay shows continues to be the eradicated value.
'Square root'
The root flow signal is used for the output signal. The 'Flow (square root)'
output signal is indicated on the on-site display with a root symbol.
'Table'The output is defined according to the scaled variable / pressure table entered.

Selection• Linear• Square root *• TablePressure value 1Navigation

 Application → Sensor → Scaled variable → P. value 1

Description

Enter pressure for the first scaling point. 'Scaled variable value 1' will be allocated to thispressure.

User entry

Signed floating-point number

Scaled variable value 1
Navigation

 Application → Sensor → Scaled variable → Sc. var.value 1

Description

Enter value for the first scaling point. This value is allocated to 'Pressure value 1'.

User interface

Signed floating-point number*

64

Visibility depends on order options or device settings

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters

Pressure value 2Navigation

 Application → Sensor → Scaled variable → P. value 2

Description

Enter pressure for the second scaling point. 'Scaled variable value 2' will be allocated to thispressure.

User entry

Signed floating-point number

Scaled variable value 2Navigation

 Application → Sensor → Scaled variable → Sc. var.value 2

Description

Enter value for the second scaling point. This value is allocated to 'Pressure value 2'.

User entry

Signed floating-point number

Low flow cut offNavigation

 Application → Sensor → Scaled variable → Low flow cut off

Description

When activated, this function suppresses small flows which can lead to large fluctuationsin the measured value.

User entry

0.0 to 50.0 %Activate table

Navigation

Application → Sensor → Scaled variable → Activate table

Selection• Disable• EnablePressure

Navigation

User entry

Signed floating-point number

Endress+Hauser

Application → Sensor → Scaled variable → Pressure

65Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL

Scaled variable

Navigation

Application → Sensor → Scaled variable → Scaled variable

User entry

Signed floating-point number

3.5.4
Navigation"PROFINET" submenu
Application → PROFINET"Configuration" submenu
Navigation

Application → PROFINET → Configuration

PROFINET device name
Navigation

 Application → PROFINET → Configuration → PROFINET DevName

Description

Shows the short form of the PROFINET device name for the measuring point

User interface

Character string comprising numbers, letters and special characters

PROFINET device name
Navigation

 Application → PROFINET → Configuration → PROFINET DevName

Description

Up to 240 characters are allowed.
The following syntax must be used:- 1 or more identifiers, separated with [.]- Identifier length is 1 to 63 characters- Identifier consists of [a-z 0-9] only lowercase letters and numbers allowed.

User entry

Character string comprising numbers, letters and special characters (240)66

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters

Parameter change acknowledge mode
Navigation

 Application → PROFINET → Configuration → ParaChngAcknMode

Description

Select how to acknowledge the displayed flag when changing the device configuration:- "Auto acknowledge": the flag disappears automatically after 20 seconds.
- "Manual acknowledge": the flag must be acknowledged manually.

Selection• Auto acknowledge• Manual acknowledge

Acknowledge parameter change
Navigation

 Application → PROFINET → Configuration → AcknParaChange

Description

If the Option "Manual acknowledge" is selected as the acknowledgement type, then aparameter change must be acknowledged with the "Reset update event flag" option.

Selection• No acknowledge• Reset update event flag

Descriptor
Navigation

 Application → PROFINET → Configuration → Descriptor

Description

Enter a description for the measuring point

User entry

Character string comprising numbers, letters and special characters (54)"Analog input 1 to 7" submenu
Navigation

Application → PROFINET → Analog input → Analog input 1 to 7

Process value
Navigation

 Application → PROFINET → Analog input → Analog input 1 to 7 → Process value

Description

Shows the process value reported to the controller for further processing

User interface

Signed floating-point number

Endress+Hauser

67Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL

Assign process variable
Navigation

 Application → PROFINET → Analog input → Analog input 1 to 7 → Assign variable

Description

Select a process variable

User interface• Pressure *• Scaled variable *• Sensor temperature *• Sensor pressure *• Electronics temperature *• Median of pressure signal *• Noise of pressure signal *Additional information

User interface"Sensor pressure" option
Sensor Pressure is the raw signal from sensor before damping and position adjustment.

Damping
Navigation

 Application → PROFINET → Analog input → Analog input 1 to 7 → Damping

Description

Enter time constant for input damping (PT1 element). Damping reduces the effect offluctuations in the measured value on the output signal.

User entry

Positive floating-point number"Binary input 1 to 2" submenu
Navigation

Application → PROFINET → Binary input → Binary input 1 to 2

Controller input value
Navigation

 Application → PROFINET → Binary input → Binary input 1 to 2 → ControllInputVal

Description

Shows for each device function the state reported to the controller for further processing

User interface

0 to 255*

68

Visibility depends on order options or device settings

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters"Binary output" submenu
Navigation

Application → PROFINET → Binary output

Set point value
Navigation

 Application → PROFINET → Binary output → Set point value

User entry

0 to 255

BO block output value
Navigation

 Application → PROFINET → Binary output → BOBlockOutValue

Description

Shows for each device function the state reported to the measuring device for further
processing

User entry

0 to 255

Failure behavior
Navigation

 Application → PROFINET → Binary output → Failure behavior

Description

Select failure behavior in the event of a failure (value with status ‘Bad’)

Selection• Fixed value• Last valid value• Actual value

Failure behavior delay
Navigation

 Application → PROFINET → Binary output → FailBehavDelay

Description

Enter a delay until in the event of a failure (value with status ‘Bad’) the failure behavior
specified applies

User entry

Positive floating-point number

Endress+Hauser

69Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL

Fixed value
Navigation

 Application → PROFINET → Binary output → Fixed value

Description

Enter value to report in the event of a failure (value with status ‘Bad’)

User entry

0 to 255"Information" submenu
Navigation

Application → PROFINET → Information

Device ID
Navigation

 Application → PROFINET → Information → Device ID

User interface

0 to 65 535

PA profile version
Navigation

 Application → PROFINET → Information → PA profile vers.

User interface

0 to 65 535"Application relation" submenu
Navigation

Application → PROFINET → Applicat. relat.

AR state
Navigation

 Application → PROFINET → Applicat. relat. → AR state

Description

Shows whether an AR connection and a system redundancy have been established

User interface• Active• Not active• Redundancy 1AR active• Redundancy 2AR active

70

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters

MAC address IO controller
Navigation

 Application → PROFINET → Applicat. relat. → MAC IO contr.

Description

Shows the MAC address of the only or of the primary IO controller

User interface

Character string comprising numbers, letters and special characters

MAC address backup IO controller
Navigation

 Application → PROFINET → Applicat. relat. → MAC backup IO c.

Description

Shows the MAC adress of the backup IO controller

User interface

Character string comprising numbers, letters and special characters

IP address IO controller
Navigation

 Application → PROFINET → Applicat. relat. → IP IO controller

Description

Shows the IP address of the only or of the primary IO controller

User interface

Character string comprising numbers, letters and special characters

IP address backup IO controller
Navigation

 Application → PROFINET → Applicat. relat. → IP backup IO c.

Description

Shows the IP adress of the backup IO controller

User interface

Character string comprising numbers, letters and special characters

Endress+Hauser

71Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL

3.6"System" menu

Navigation

3.6.1



System"Device management" submenu

Navigation



System → Device manag.

Device tag
Navigation

 System → Device manag. → Device tag

Description

Enter a name for the measuring point to identify the measuring device in the plantUser entry

Character string comprising numbers, letters and special characters (32)

Locking status
Navigation

 System → Device manag. → Locking status

Description

Displays the active write protection.

User interface• Hardware locked• Temporarily locked

Additional information

User interface
If two or more types of write protection are active, the write protection with the highest
priority is shown on the local display. In the operating tool all active types of write
protection are displayed.
Detailed information on access authorization is provided in the "User roles andassociated access authorization" and "Operating concept" sections of the Operations
Instructions for the device.
Selection
Function scope of the "Locking status" parameter72

Options

Description

None

The access status displayed in the Access status display parameter applies. Only
appears on local display.

Hardware locked

The DIP switch for hardware locking is activated on the main electronics module.
This prevents write access to the parameters (e.g. via the local display or operatingtool).

Temporarily locked

Write access to the parameters is temporarily locked due to device-internal
processing (e.g. data upload/download, reset). Once the internal processing has
been completed, the parameters can be changed once again.

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters

Configuration counter
Navigation

 System → Device manag. → Config. counter

Description

Shows the number of changes made to static parameters (e.g. configuration parameters)

User interface

0 to 65 535

Reset device

Navigation

 System → Device manag. → Reset device

Description

Use this function to choose whether to reset the device configuration - either entirely or inpart - to a defined state.

Selection• Cancel• To factory defaults *• To delivery settings *• Restart device

Additional information

Selection
Options

Description

CancelNo action is executed and the user exits the parameter.

To factory defaults

Every parameter is reset to its factory setting.

To delivery settings

Every parameter for which a customer-specific default setting was ordered is resetto this customer-specific value. All other parameters are reset to the factory
setting.

 This option is not visible if no customer-specific settings have been ordered.
Restart device*

The restart resets every parameter whose data are in the volatile memory (RAM)to the factory setting (e.g. measured value data). The device configuration remains
unchanged.

Visibility depends on order options or device settings

Endress+Hauser

73Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL

3.6.2"User management" submenu

Navigation



System → User manag.

User role
Navigation

 System → User manag. → User role

Description

Shows the access authorization to the parameters via the operating tool

User interface• Operator• Maintenance• Expert• Production• Development

Password
Navigation



System → User manag. → Password

Description

Enter the password for the 'Maintenance' user role to get access to the functionality of thisrole.

User entry

Character string comprising numbers, letters and special characters (16)Enter access code

Navigation



System → User manag. → Ent. access code

Description

Use this function to enter the user-specific release code to remove parameter write
protection in the operating tool.

User entry

0 to 9 999

Status password entry
Navigation

 System → User manag. → Status pw entry

Description

Use this function to display the status of the password verification.

User interface• -------• Wrong password• Password rule violated

74

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters• Password accepted• Permission denied• Confirm PW mismatch• Reset password accepted• Invalid user role
• Wrong sequence of entry

New passwordNavigation

 System → User manag. → New password

Description

Define the new 'Maintenance' password.
A new password is valid after it has been confirmed within the 'Confirm new password'
parameter.
Any valid password consists of 4 to 16 characters and can contain letters and numbers.

User entry

Character string comprising numbers, letters and special characters (16)

Confirm new passwordNavigation

 System → User manag. → Conf. new passw.

Description

Enter the new password again to confirm.

User entry

Character string comprising numbers, letters and special characters (16)

Old passwordNavigation

 System → User manag. → Old password

Description

Enter the current password, to subsequently change the existing password.

User entry

Character string comprising numbers, letters and special characters (16)Reset password
NavigationDescription

Enter a code to reset the current 'Maintenance' password.
The code is deliverd by your local support.

User entry

Character string comprising numbers, letters and special characters (16)Endress+Hauser

System → User manag. → Reset password

75Description of device parameters

3.6.3
Navigation

Deltabar PMD55B PROFINET with Ethernet-APL"Connectivity" submenu


System → Connectivity"Interfaces" submenu
Navigation



System → Connectivity → Interfaces

Display operation



Navigation

 System → Connectivity → Interfaces → DisplayOperation

Selection• Disable• Enable

Web server functionality



Navigation

 System → Connectivity → Interfaces → Webserver funct.

Description

Switch web server on and off, switch off HTML.

Selection• Disable• Enable

Bluetooth activation
Navigation

 System → Connectivity → Interfaces → Bluetooth active

Description

If Bluetooth is deactivated, it can only be reactivated via the display or the operating tool.
Reactivating via the SmartBlue app is not possible.

Selection• Disable• Enable

Service (UART-CDI)Navigation

 System → Connectivity → Interfaces → Service (CDI)

Selection• Disable• Enable

76

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters"Ethernet" submenu
Navigation



System → Connectivity → Ethernet

MAC address
Navigation

 System → Connectivity → Ethernet → MAC Address

Description

Shows the MAC address of the measuring device

User interface

Character string comprising numbers, letters and special characters

IP addressNavigation



System → Connectivity → Ethernet → IP address

Description

Enter the IP address of the measuring device

User entry

Character string comprising numbers, letters and special characters (15)Subnet mask

Navigation



System → Connectivity → Ethernet → Subnet mask

Description

Enter subnet mask of the measuring device

User entry

Character string comprising numbers, letters and special characters (15)Default gateway

NavigationDescription

Enter IP address for the default gateway of the measuring device

User entry

Character string comprising numbers, letters and special characters (15)Endress+Hauser

System → Connectivity → Ethernet → Default gateway

77Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL

Service IP active
Navigation

 System → Connectivity → Ethernet → Service IP act.

User interface

• No
• Yes

Interface connection status
Navigation

 System → Connectivity → Ethernet → Interface status

User interface• Connected• Not connectedInterface speed
Navigation

 System → Connectivity → Ethernet → Interface speed

User interface

Positive integer

Duplex status
Navigation

 System → Connectivity → Ethernet → Duplex status

User interface• Full duplex• Half duplex• Unknown

Auto negotiation status
Navigation

 System → Connectivity → Ethernet → Auto negot.stat.

User interface• Idle• In progress• Completed• Failed• Speed detection failed

78

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters

Received packet number
Navigation

 System → Connectivity → Ethernet → RX packet no.

User interface

Positive integer

Sent packet number
Navigation

 System → Connectivity → Ethernet → TX packet number

User interface

Positive integer

Number of failed received packets
Navigation

 System → Connectivity → Ethernet → FailRcvdPackets

User interface

Positive integer

Number of failed sent packets
Navigation

 System → Connectivity → Ethernet → FailTXPacketsNo.

User interface

Positive integer

Signal to noise ratio
Navigation

 System → Connectivity → Ethernet → SNR

Description

Shows the signal to noise ratio of the Ethernet-APL connection. A value >21dB is goodand >23dB is excellent.

User interface

Signed floating-point number

Number of failed received packets
Navigation

 System → Connectivity → Ethernet → FailRXPacketsNo.

Description

Shows the number of failed received packets.

Endress+Hauser

79Description of device parameters

User interface

Deltabar PMD55B PROFINET with Ethernet-APL

0 to 65 535

Active TCP connections
Navigation

 System → Connectivity → Ethernet → Act. TCP connec.

User interface

0 to 65 535

Supported TCP connections
Navigation

 System → Connectivity → Ethernet → Supported TCPUser interface

0 to 65 535

TCP connection requests
Navigation

 System → Connectivity → Ethernet → TCPConnecRequest

User interface

0 to 65 535

TCP connection timeouts
Navigation

 System → Connectivity → Ethernet → TCPConnecTimeout

User interface

0 to 255

Number of TCP connections closed
Navigation

 System → Connectivity → Ethernet → TCPConnect.close

User interface

0 to 255

Number of received TCP packets
Navigation

 System → Connectivity → Ethernet → No.RX TCP Packet

User interface

Positive integer

80

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters

TCP sent packet number
Navigation

 System → Connectivity → Ethernet → TCP TX PacketNo.

User interface

Positive integer

Number of TCP failed received packets
Navigation

 System → Connectivity → Ethernet → TCPFailRXPackets

User interface

Positive integer

Available UDP ports
Navigation

 System → Connectivity → Ethernet → Avail. UDP ports

User interface

Positive integer

UDP received packet number
Navigation

 System → Connectivity → Ethernet → UDP RX PacketNo.

User interface

Positive integer

UDP sent packet number
Navigation

 System → Connectivity → Ethernet → UDP TX PacketNo.

User interface

Positive integer

Number of UDP failed received packets
Navigation

 System → Connectivity → Ethernet → UDPFailRXPackets

User interface

Positive integer

Endress+Hauser

81Description of device parameters

3.6.4

Deltabar PMD55B PROFINET with Ethernet-APL"Display" submenu

Navigation



System → Display

Language
Navigation

 System → Display → Language

Prerequisite

A local display is provided.

Description

Use this function to select the configured language on the local display.

Selection• English• Deutsch• Français• Español• Italiano• Nederlands• Portuguesa• Polski• русский язык (Russian)• Svenska• Türkçe• 中文 (Chinese)• 日本語 (Japanese)• 한국어 (Korean)• Bahasa Indonesia• tiếng Việt (Vietnamese)• čeština (Czech)

Factory setting

English (alternatively, the ordered language is preset in the device)

Format display
Navigation

 System → Display → Format display

Prerequisite

A local display is provided.

Description

Use this function to select how the measured value is shown on the local display.

Selection• 1 value, max. size
• 2 values

82

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Additional information

Description of device parameters

Description
The display format (size, bar graph etc.) and number of measured values displayed
simultaneously (1 to 4) can be configured. This setting only applies to normal operation.
• The Value 1 display parameter (→  83)...Value 8 display parameter Value 4
display parameter (→  84) are used to specify which measured values areshown on the local display and in what order.
• If more measured values are specified than the display mode selected permits, thenthe values alternate on the device display. The display time until the next change isconfigured using the Display interval parameter.

Value 1 display

Navigation

 System → Display → Value 1 display

Prerequisite

A local display is provided.

Description

Use this function to select one of the measured values shown on the local display.

Selection• Pressure• Scaled variable• Sensor temperature

Additional information

Description
If several measured values are displayed at once, the measured value selected here will be
the first value to be displayed. The value is only displayed during normal operation.
The Format display parameter (→  82) is used to specify how many measured
values are displayed simultaneously and how.
Dependency
The unit of the displayed measured value is taken from the System units submenu.

Value 2 display

Navigation

 System → Display → Value 2 display

Prerequisite

A local display is provided.

Description

Use this function to select one of the measured values shown on the local display.

Selection• None• Pressure• Scaled variable• Sensor temperature

Endress+Hauser

83Description of device parameters

Additional information

Deltabar PMD55B PROFINET with Ethernet-APL

Description
If several measured values are displayed at once, the measured value selected here will be
the second value to be displayed. The value is only displayed during normal operation.
The Format display parameter (→  82) is used to specify how many measured
values are displayed simultaneously and how.
Dependency
The unit of the displayed measured value is taken from the System units submenu.

Value 3 display

Navigation

 System → Display → Value 3 display

Prerequisite

A local display is provided.

Description

Use this function to select one of the measured values shown on the local display.

Selection• None• Pressure• Scaled variable• Sensor temperature

Additional information

Description
If several measured values are displayed at once, the measured value selected here will be
the third value to be displayed. The value is only displayed during normal operation.
The Format display parameter (→  82) is used to specify how many measured
values are displayed simultaneously and how.
Selection
The unit of the displayed measured value is taken from the System units submenu.

Value 4 display

Navigation

 System → Display → Value 4 display

Prerequisite

A local display is provided.

Description

Use this function to select one of the measured values shown on the local display.

Selection• None• Pressure• Scaled variable• Sensor temperature

84

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Additional information

Description of device parameters

Description
If several measured values are displayed at once, the measured value selected here will be
the fourth value to be displayed. The value is only displayed during normal operation.
The Format display parameter (→  82) is used to specify how many measured
values are displayed simultaneously and how.
Selection
The unit of the displayed measured value is taken from the System units submenu.

Contrast display
Navigation

 System → Display → Contrast display

Description

Adjust local display contrast setting to ambient conditions (e.g. lighting or reading angle)

User entry

20 to 80 %Factory setting

Depends on the display

Additional information

Set the contrast via the push-buttons:• Weaker: Press the and buttons simultaneously• Stronger: Press the and buttons simultaneously
E

E

3.6.5
Navigation"Date/time" submenu


System → Date/time

Date/time
Navigation

 System → Date/time → Date/time

Description

Displays the date and time entered.

User interface

Character string comprising numbers, letters and special characters

Time zoneNavigation

 System → Date/time → Time zone

Description

Select the time zone. Every time the time zone is changed, a logbook entry is created.

Endress+Hauser

85Description of device parameters

Selection

Deltabar PMD55B PROFINET with Ethernet-APL

Other units• UTC-12:00• UTC-11:00• UTC-10:00• UTC-09:30• UTC-09:00• UTC-08:00• UTC-07:00• UTC-06:00• UTC-05:00• UTC-04:00• UTC-03:30• UTC-03:00• UTC-02:30• UTC-02:00• UTC-01:00• UTC 00:00• UTC+01:00• UTC+02:00• UTC+03:00• UTC+03:30• UTC+04:00• UTC+04:30• UTC+05:00• UTC+05:30• UTC+05:45• UTC+06:00• UTC+06:30• UTC+07:00• UTC+08:00• UTC+08:45• UTC+09:00• UTC+09:30• UTC+10:00• UTC+10:30• UTC+11:00• UTC+12:00• UTC+12:45• UTC+13:00• UTC+13:45• UTC+14:00

Enable NTPNavigation

Selection

• No
• Yes

86

System → Date/time → Enable NTP

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters

NTP server addressNavigation



System → Date/time → NTP server add.

Description

IP address of the NTP server.

User entry

Character string comprising numbers, letters and special characters (64)Clock synchronized
Navigation

 System → Date/time → Clock synch.

Description

Timestamp of last synchronization with an NTP server.

User interface

Character string comprising numbers, letters and special characters

3.6.6"Geolocation" submenu

Navigation



System → Geolocation

Location description



Navigation

 System → Geolocation → Location descr.

Description

Enter a description for the location

User entry

Character string comprising numbers, letters and special characters (32)Longitude

Navigation

 System → Geolocation → Longitude

Description

Enter the longitude.

User entry–180 to 180 °Endress+Hauser

87Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL

LatitudeNavigation

 System → Geolocation → Latitude

Description

Enter latitude

User entry–90 to 90 °AltitudeNavigation

 System → Geolocation → Altitude

Description

Enter altitude

User entry

Signed floating-point number

3.6.7
Navigation"Information" submenu


System → Information

Device name
Navigation

 System → Information → Device name

Description

Displays the name of the transmitter. It can also be found on the nameplate of thetransmitter.

User interface

Max. 32 characters such as letters or numbers.

Manufacturer
Navigation

 System → Information → Manufacturer

User interface

Character string comprising numbers, letters and special characters

88

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters

Serial number
Navigation

 System → Information → Serial number

Description

Displays the serial number of the measuring device.
The number can be found on the nameplate of the sensor and transmitter.

User interface

Max. 11-digit character string comprising letters and numbers.

Additional information

Description
Uses of the serial number• To identify the measuring device quickly, e.g. when contacting Endress+Hauser.
• To obtain specific information on the measuring device using the Device Viewer:
www.endress.com/deviceviewer

Order code

Navigation

 System → Information → Order code

Description

Shows the device order code.

User interface

Character string composed of letters, numbers and certain punctuation marks (e.g. /).

Factory setting–

Additional information

Description
The order code is generated from the extended order code through a process of reversible
transformation. The extended order code indicates the attributes for all the device featuresin the product structure. The device features are not directly readable from the order code.
Uses of the order code
• To order an identical spare device.
• To identify the device quickly and easily, e.g. when contacting Endress+Hauser.

Firmware version
Navigation

 System → Information → Firmware version

Description

Displays the device firmware version that is installed.

User interface

Character string in the format xx.yy.zz

Additional information

User interface
The Firmware version is also located:• On the title page of the Operating instructions• On the transmitter nameplate

Endress+Hauser

89Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL

Hardware version
Navigation

 System → Information → Hardware version

User interface

Character string comprising numbers, letters and special characters

Extended order code 1Navigation



System → Information → Ext. order cd. 1

Description

The extended order code is an alphanumeric code containing all information to identify
the device and its options.

User interface

Character string

Factory setting–

Additional information

Description
The extended order code indicates the version of all the features of the product structure
for the measuring device and thus uniquely identifies the measuring device.

Extended order code 2Navigation



System → Information → Ext. order cd. 2

Description

The extended order code is an alphanumeric code containing all information to identify
the device and its options.
The extended order code can also be found on the nameplate of the sensor andtransmitter in the "Ext. ord. cd." field.

User interface

Character string

Factory setting–

Extended order code 3

Navigation



System → Information → Ext. order cd. 3

Description

The extended order code is an alphanumeric code containing all information to identify
the device and its options.
The extended order code can also be found on the nameplate of the sensor andtransmitter in the "Ext. ord. cd." field.

User interface
90

Character string
Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Factory setting

Description of device parameters–

Checksum
Navigation

 System → Information → Checksum

Description

Checksum for Firmware version.

User interface

Positive integer

3.6.8
Navigation"Software configuration" submenu


System → Softw. config.

Activate SW option



Navigation

 System → Softw. config. → Activate SW opt.

Description

Use this function to enter an activation code to enable an additional, ordered software
option.

User entry

Max. 10-digit string of numbers.

Factory setting

Depends on the software option ordered

Additional information

Description
If a measuring device was ordered with an additional software option, the activation codeis programmed in the device at the factory.
User entry
To activate a software option subsequently, please contact your Endress+Hauser salesorganization.
NOTE!
The activation code is linked to the serial number of the measuring device and varies
according to the device and software option.
If an incorrect or invalid code is entered, this results in the loss of software options thathave already been activated.
▸ Before you enter a new activation code, make a note of the current activation code fromthe parameter protocol.
▸ Enter the new activation code provided by Endress+Hauser when the new software
option was ordered.
▸ If the code entered is incorrect or invalid, enter the old activation code from theparameter protocol.

Endress+Hauser

91Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL▸ Have the Endress+Hauser sales organization check the new activation code
remembering to specify the serial number or ask for the code again.
Example for a software option
Order code for "Application package", option EA "Extended HistoROM"

Software option overview
Navigation

 System → Softw. config. → SW option overv.

Description

Shows all enabled software options

User interface• Heartbeat Verification• Heartbeat Monitoring"Firmware update" wizard
Depending on device and software interface, it is possible to update main software, sensor
and display firmware.
Be sure to have a valid firmware package (e.g.:
Device_Name_BusType_Vxx.yy.zz_bbbb_pkgVuu.vv.ww.sfu) already available on yoursystem.
A list of available firmware can be found on “www.endress.com” or asking to local service
supplier.
If the firmware is validated, a further confirmation will be prompted before to start the
update process.
It is always possible to cancel the firmware update process before final confirmation.
Navigation



System → Softw. config. → Firmware update"Start update" wizard
Warning: Firmware update is irreversible. It is not always possible to downgrade thefirmware after a successfully update to ensure compatibility.
Be sure to save configuration before to start the firmware update process.
After firmware confirmation a device restart can occur; be sure to consider all the related
safety measures.

92

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Navigation

Description of device parameters

System → Softw. config. → Firmware update → Start update

I have read the warning texts.
Navigation

Selection

Yes
System → Softw. config. → Firmware update → Start update → warning texts"Device information" wizard
Warning: Firmware update is irreversible. It is not always possible to downgrade thefirmware after a successfully update to ensure compatibility.
Be sure to save configuration before to start the firmware update process.
After firmware confirmation a device restart can occur; be sure to consider all the related
safety measures.
Navigation



System → Softw. config. → Firmware update → Device info

Device tag
Navigation



System → Softw. config. → Firmware update → Device info → Device tag

Description

Displays a unique name for the measuring point so it can be identified quickly within theplant. The name is displayed in the header.

User interface

Max. 32 characters, such as letters, numbers or special characters (e.g. @, %, /).

Additional information

User interface
1

XXXXXXXXX

A0029422

1

Position of the header text on the display

The number of characters displayed depends on the characters used.

Endress+Hauser

93Description of device parameters

Deltabar PMD55B PROFINET with Ethernet-APL

Device name
Navigation



System → Softw. config. → Firmware update → Device info → Device name

Description

Displays the name of the transmitter. It can also be found on the nameplate of thetransmitter.

User interface

Max. 32 characters such as letters or numbers.

Package version
Navigation



System → Softw. config. → Firmware update → Device info → Package version

Description

Current version of installed firmware package.
The package is an ‘.sfu’ extension file containing
all needed device software components.

User interface

Positive integer"Select file" wizard
Select firmware update file to be transferred to the device.
Navigation



System → Softw. config. → Firmware update → Select file

File check status
Navigation

User interface• Active• Failed• Not done
• Passed

94

System → Softw. config. → Firmware update → Select file → File check st.

Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Description of device parameters"Performing verification" wizard
Navigation



System → Softw. config. → Firmware update → Perform.verific.

File validation status
Navigation



System → Softw. config. → Firmware update → Perform.verific. → File val. st.

User interface• Active• Failed• Not done
• Passed"Finish" wizard
Caution: Firmware updates are irreversible.
Firmware update has been transferred successfully. Continue to start update.
Navigation



System → Softw. config. → Firmware update → Finish

I have read the warning texts.
Navigation

Selection

Yes

Endress+Hauser


System → Softw. config. → Firmware update → Finish → warning texts

95IndexDeltabar PMD55B PROFINET with Ethernet-APL

Index
0…9

436 Diagnostic behavior (Parameter) . . . . . . . . . . . . . 48
436 Event category (Parameter) . . . . . . . . . . . . . . . . 49
500 Diagnostic behavior (Parameter) . . . . . . . . . . . . . 49
500 Event category (Parameter) . . . . . . . . . . . . . . . . 49
500 Process alert pressure (Parameter) . . . . . . . . . . . 46
501 Diagnostic behavior (Parameter) . . . . . . . . . . . . . 50
501 Event category (Parameter) . . . . . . . . . . . . . . . . 50
501 Process alert scaled variable (Parameter) . . . . . . . 47
502 Diagnostic behavior (Parameter) . . . . . . . . . . . . . 50
502 Event category (Parameter) . . . . . . . . . . . . . . . . 51
841 Event category (Parameter) . . . . . . . . . . . . . . . . 51
900 Diagnostic behavior (Parameter) . . . . . . . . . . . . . 52
900 Event category (Parameter) . . . . . . . . . . . . . . . . 52
906 Diagnostic behavior (Parameter) . . . . . . . . . . . . . 52
906 Event category (Parameter) . . . . . . . . . . . . . . . . 53

A

Acknowledge parameter change (Parameter) . . . . . . . 67
Activate SW option (Parameter) . . . . . . . . . . . . . . . . 91
Activate table (Parameter) . . . . . . . . . . . . . . . . . . . . . 65
Active diagnostics (Parameter) . . . . . . . . . . . . . . . . . 38
Active diagnostics (Submenu) . . . . . . . . . . . . . . . . . . 38
Active TCP connections (Parameter) . . . . . . . . . . . . . 80
Altitude (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . . 88
Analog input 1 to 7 (Submenu) . . . . . . . . . . . . . . . . . 67
Application (Menu) . . . . . . . . . . . . . . . . . . . . . . . . . . 54
Application relation (Submenu) . . . . . . . . . . . . . . . . . 70
AR state (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . . 70
Assign process variable (Parameter) . . . . . . . . . . . 37, 68
Assign scaled variable? (Parameter) . . . . . . . . . . . . . . 29
Auto negotiation status (Parameter) . . . . . . . . . . . . . 78
Available UDP ports (Parameter) . . . . . . . . . . . . . . . . 81

B

Binary input 1 to 2 (Submenu) . . . . . . . . . . . . . . . . . . 68
Binary output (Submenu) . . . . . . . . . . . . . . . . . . . . . 69
Bluetooth activation (Parameter) . . . . . . . . . . . . . . . . 76
BO block output value (Parameter) . . . . . . . . . . . . . . . 69

C

Calibration offset (Parameter) . . . . . . . . . . . . . . . . . . 58
Checksum (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 91
Clock synchronized (Parameter) . . . . . . . . . . . . . . . . . 87
Commissioning (Wizard) . . . . . . . . . . . . . . . . . . . . . . 23
Configuration (Submenu) . . . . . . . . . . . . . . . . . . . 48, 66
Configuration counter (Parameter) . . . . . . . . . . . . . . 73
Confirm new password (Parameter) . . . . . . . . . . . . . . 75
Connectivity (Submenu) . . . . . . . . . . . . . . . . . . . . . . 76
Contrast display (Parameter) . . . . . . . . . . . . . . . . . . . 85
Controller input value (Parameter) . . . . . . . . . . . . . . . 68
Counter limit overruns sensor Pmax (Parameter) . . . . 43
Counter limit overruns sensor Tmax (Parameter) . . . . 44
Counter limit underruns sensor Pmin (Parameter) . . . 41
Counter limit underruns sensor Tmin (Parameter) . . . 42
Counter overruns of user limit Pmax (Parameter) . . . . 43
96

Counter overruns of user limit Tmax (Parameter) . . . . 44
Counter underruns of user limit Pmin (Parameter) . . . 41
Counter underruns of user limit Tmin (Parameter) . . . 42

D

Damping (Parameter) . . . . . . . . . . . . . . . . . . 29, 60, 68
Date/time (Parameter) . . . . . . . . . . . . . . . . . . . . 27, 85
Date/time (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 85
Decimal places pressure (Parameter) . . . . . . . . . . . . . 55
Decimal places scaled variable (Parameter) . . . . . . . . 57
Default gateway (Parameter) . . . . . . . . . . . . . . . . . . . 77
Description of device parameters . . . . . . . . . . . . . . . . 19
Descriptor (Parameter) . . . . . . . . . . . . . . . . . . . . 28, 67
Device ID (Parameter) . . . . . . . . . . . . . . . . . . . . . 28, 70
Device identification (Wizard) . . . . . . . . . 23, 25, 26, 27
Device information (Menu) . . . . . . . . . . . . . . . . . . . . 22
Device information (Wizard) . . . . . . . . . . . . . . . . . . . 93
Device management (Submenu) . . . . . . . . . . . . . . . . . 72
Device name (Parameter) . . . . . . . . . . . . . . . . 23, 88, 94
Device tag (Parameter) . . . . . . . . . . . . . . . . . 23, 72, 93
Diagnostic event simulation (Parameter) . . . . . . . . . . 45
Diagnostic settings (Submenu) . . . . . . . . . . . . . . . . . . 46
Diagnostics (Menu) . . . . . . . . . . . . . . . . . . . . . . . . . . 38
Display (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . 82
Display operation (Parameter) . . . . . . . . . . . . . . . . . . 76
Do not show this message again (Parameter) . . . . . . . 22
Document
Explanation of the structure of a parameter
description . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Function . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Structure . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Symbols used . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 5
Target group . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Using the document . . . . . . . . . . . . . . . . . . . . . . . . 4
Document function . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Duplex status (Parameter) . . . . . . . . . . . . . . . . . . . . . 78

EElectronics temperature (Parameter) . . . . . . . . . . . . . 54
Enable NTP (Parameter) . . . . . . . . . . . . . . . . . . . . . . 86
Enter access code (Parameter) . . . . . . . . . . . . . . . . . . 74
Ethernet (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . 77
Event logbook (Submenu) . . . . . . . . . . . . . . . . . . . . . 40
Extended order code 1 (Parameter) . . . . . . . . . . . 24, 90
Extended order code 2 (Parameter) . . . . . . . . . . . 24, 90
Extended order code 3 (Parameter) . . . . . . . . . . . 24, 90

F

Failure behavior (Parameter) . . . . . . . . . . . . . . . . . . . 69
Failure behavior delay (Parameter) . . . . . . . . . . . . . . 69
File check status (Parameter) . . . . . . . . . . . . . . . . . . . 94
File validation status (Parameter) . . . . . . . . . . . . . . . 95
Filter options (Parameter) . . . . . . . . . . . . . . . . . . . . . 40
Finish (Wizard) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 95
Firmware update (Wizard) . . . . . . . . . . . . . . . . . . . . 92
Firmware version (Parameter) . . . . . . . . . . . . . . . . . . 89
Fixed value (Parameter) . . . . . . . . . . . . . . . . . . . . . . 70
Endress+Hauser

Deltabar PMD55B PROFINET with Ethernet-APL

Format display (Parameter) . . . . . . . . . . . . . . . . . . . . 82
Free text (Parameter) . . . . . . . . . . . . . . . . . . 31, 57, 63
Function
see Parameter

G

Geolocation (Submenu) . . . . . . . . . . . . . . . . . . . . . . . 87
Guidance (Menu) . . . . . . . . . . . . . . . . . . . . . . . . . 19, 23

H

Hardware version (Parameter) . . . . . . . . . . . . . . . . . . 90
High alert value (Parameter) . . . . . . . . . . . . . . . . 47, 48
HP/LP swap (Parameter) . . . . . . . . . . . . . . . . . . . . . . 60

I

I have read the warning texts. (Parameter) . . . . . . 93, 95
Information (Submenu) . . . . . . . . . . . . . . . . . . . . 70, 88
Interface connection status (Parameter) . . . . . . . . . . . 78
Interface speed (Parameter) . . . . . . . . . . . . . . . . . . . . 78
Interfaces (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 76
IP address (Parameter) . . . . . . . . . . . . . . . . . . . . 27, 77
IP address backup IO controller (Parameter) . . . . . . . . 71
IP address IO controller (Parameter) . . . . . . . . . . . . . . 71

L

Language (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 82
Latitude (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . . 88
Location description (Parameter) . . . . . . . . . . . . . . . . 87
Locking status (Parameter) . . . . . . . . . . . . . . . . . 25, 72
Longitude (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 87
Low alert value (Parameter) . . . . . . . . . . . . . . 46, 47, 48
Low flow cut off (Parameter) . . . . . . . . . . . . . . . . 33, 65
Lower Range Limit (Parameter) . . . . . . . . . . . 34, 36, 61
Lower range value (Parameter) . . . . . . . . . . . . . . . . . 59
Lower sensor trim (Parameter) . . . . . . . . . . . . . . . . . 59

M

MAC address (Parameter) . . . . . . . . . . . . . . . . . . 28, 77
MAC address backup IO controller (Parameter) . . . . . 71
MAC address IO controller (Parameter) . . . . . . . . . . . 71
Manufacturer (Parameter) . . . . . . . . . . . . . . . . . . . . . 88
Manufacturer ID (Parameter) . . . . . . . . . . . . . . . . . . . 28
Maximum electronics temperature (Parameter) . . . . . 44
Maximum sensor temperature (Parameter) . . . . . . . . 43
Maximum terminal voltage (Parameter) . . . . . . . . . . 44
Measured values (Submenu) . . . . . . . . . . . . . . . . . . . 54
Measurement adjustments (Wizard) . . . . . . . 29, 30, 32
Measuring units (Submenu) . . . . . . . . . . . . . . . . . . . . 55
Menu
Application . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 54
Device information . . . . . . . . . . . . . . . . . . . . . . . . 22
Diagnostics . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 38
Guidance . . . . . . . . . . . . . . . . . . . . . . . . . . . . 19, 23
System . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 72
Minimum electronics temperature (Parameter) . . . . . 42
Minimum sensor temperature (Parameter) . . . . . . . . 41
Minimum span (Parameter) . . . . . . . . . . . . . . 34, 37, 61
Minimum terminal voltage (Parameter) . . . . . . . . . . . 42
Minimum/maximum values (Submenu) . . . . . . . . . . . 41

Endress+Hauser

Index

N

New password (Parameter) . . . . . . . . . . . . . . . . . . . . 75
NTP server address (Parameter) . . . . . . . . . . . . . . . . . 87
Number of failed received packets (Parameter) . . . . . . 79
Number of failed sent packets (Parameter) . . . . . . . . . 79
Number of received TCP packets (Parameter) . . . . . . . 80
Number of TCP connections closed (Parameter) . . . . . 80
Number of TCP failed received packets (Parameter) . . 81
Number of UDP failed received packets (Parameter) . . 81

O

Old password (Parameter) . . . . . . . . . . . . . . . . . . . . . 75
Operating time (Parameter) . . . . . . . . . . . . . . . . . . . . 40
Operating time from restart (Parameter) . . . . . . . . . . 39
Order code (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 89
Output settings (Wizard) . . . . . . . . . . . . . 33, 34, 35, 37

P

PA profile version (Parameter) . . . . . . . . . . . . . . . . . . 70
Package version (Parameter) . . . . . . . . . . . . . . . . . . . 94
Parameter
Structure of a parameter description . . . . . . . . . . . . 4
Parameter change acknowledge mode (Parameter) . . 67
Password (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 74
Performing verification (Wizard) . . . . . . . . . . . . . . . . 95
Pressure (Parameter) . . . . . . . . . . 22, 33, 34, 54, 64, 65
Pressure max (Parameter) . . . . . . . . . . . . . . . . . . . . . 43
Pressure min (Parameter) . . . . . . . . . . . . . . . . . . . . . 41
Pressure unit (Parameter) . . . . . . . . . . . . . . . 29, 30, 55
Pressure value 1 (Parameter) . . . . . . . . . . . . . . . . 35, 64
Pressure value 2 (Parameter) . . . . . . . . . . . . . . . . 36, 65
Previous diagnostics (Parameter) . . . . . . . . . . . . . . . . 39
Process (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . 51
Process value (Parameter) . . . . . . . . . . . . . . . . . . . . . 67
PROFINET (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 66
PROFINET device name (Parameter) . . . . . . . . . . 27, 66
Properties (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 46

R

Received packet number (Parameter) . . . . . . . . . . . . . 79
Reset device (Parameter) . . . . . . . . . . . . . . . . . . . . . . 73
Reset password (Parameter) . . . . . . . . . . . . . . . . . . . 75
Reset user defined counters P and T (Parameter) . . . . 42

S

Scaled variable (Parameter) . . . . . . . . . . . 22, 35, 54, 66
Scaled variable (Submenu) . . . . . . . . . . . . . . . . . . . . . 62
Scaled variable transfer function (Parameter) 33, 35, 64
Scaled variable unit (Parameter) . . . . . . . . . . . 31, 56, 62
Scaled variable value 1 (Parameter) . . . . . . . . . . . 36, 64
Scaled variable value 2 (Parameter) . . . . . . . . . . . 36, 65
Select file (Wizard) . . . . . . . . . . . . . . . . . . . . . . . . . . 94
Sensor calibration (Submenu) . . . . . . . . . . . . . . . . . . 58
Sensor configuration (Submenu) . . . . . . . . . . . . . . . . 60
Sensor limits (Submenu) . . . . . . . . . . . . . . . . . . . . . . 61
Sensor pressure range behavior (Parameter) . . . . . . . 51
Sensor temperature (Parameter) . . . . . . . . . . . . . . . . 54
Sensor temperature lower range limit (Parameter) . . . 61
Sensor temperature upper range limit (Parameter) . . . 62
97

IndexSensor Trim Reset (Parameter) . . . . . . . . . . . . . . . . . 58
Sent packet number (Parameter) . . . . . . . . . . . . . . . . 79
Serial number (Parameter) . . . . . . . . . . . . . . . . . . 23, 89
Service (UART-CDI) (Parameter) . . . . . . . . . . . . . . . . 76
Service IP active (Parameter) . . . . . . . . . . . . . . . . . . . 78
Set point value (Parameter) . . . . . . . . . . . . . . . . . . . . 69
Signal to noise ratio (Parameter) . . . . . . . . . . . . . . . . 79
Simulation (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 45
Simulation (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 45
Software configuration (Submenu) . . . . . . . . . . . . . . . 91
Software option overview (Parameter) . . . . . . . . . . . . 92
SSD Monitoring delay time (Parameter) . . . . . . . . . . . 46
SSD Out of range delay time (Parameter) . . . . . . . . . . 46
Start update (Wizard) . . . . . . . . . . . . . . . . . . . . . . . . 92
Status password entry (Parameter) . . . . . . . . . . . . . . 74
Status signal (Parameter) . . . . . . . . . . . . . . . . . . . . . 22
Submenu
Active diagnostics . . . . . . . . . . . . . . . . . . . . . . . . 38
Analog input 1 to 7 . . . . . . . . . . . . . . . . . . . . . . . 67
Application relation . . . . . . . . . . . . . . . . . . . . . . . 70
Binary input 1 to 2 . . . . . . . . . . . . . . . . . . . . . . . . 68
Binary output . . . . . . . . . . . . . . . . . . . . . . . . . . . . 69
Configuration . . . . . . . . . . . . . . . . . . . . . . . . . 48, 66
Connectivity . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 76
Date/time . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 85
Device management . . . . . . . . . . . . . . . . . . . . . . . 72
Diagnostic settings . . . . . . . . . . . . . . . . . . . . . . . . 46
Display . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 82
Ethernet . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 77
Event logbook . . . . . . . . . . . . . . . . . . . . . . . . . . . 40
Geolocation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 87
Information . . . . . . . . . . . . . . . . . . . . . . . . . . 70, 88
Interfaces . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 76
Measured values . . . . . . . . . . . . . . . . . . . . . . . . . 54
Measuring units . . . . . . . . . . . . . . . . . . . . . . . . . . 55
Minimum/maximum values . . . . . . . . . . . . . . . . . 41
Process . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 51
PROFINET . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 66
Properties . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 46
Scaled variable . . . . . . . . . . . . . . . . . . . . . . . . . . . 62
Sensor calibration . . . . . . . . . . . . . . . . . . . . . . . . . 58
Sensor configuration . . . . . . . . . . . . . . . . . . . . . . 60
Sensor limits . . . . . . . . . . . . . . . . . . . . . . . . . . . . 61
Simulation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 45
Software configuration . . . . . . . . . . . . . . . . . . . . . 91
User management . . . . . . . . . . . . . . . . . . . . . . . . 74
Subnet mask (Parameter) . . . . . . . . . . . . . . . . . . . . . 77
Supported TCP connections (Parameter) . . . . . . . . . . . 80
System (Menu) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 72

Deltabar PMD55B PROFINET with Ethernet-APL

U

UDP received packet number (Parameter) . . . . . . . . . 81
UDP sent packet number (Parameter) . . . . . . . . . . . . 81
Upper Range Limit (Parameter) . . . . . . . . . . . 34, 36, 61
Upper range value (Parameter) . . . . . . . . . . . . . . . . . 59
Upper sensor trim (Parameter) . . . . . . . . . . . . . . . . . 59
User management (Submenu) . . . . . . . . . . . . . . . . . . 74
User role (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 74
User temperature process alert (Parameter) . . . . . . . . 48

V

Value 1 display (Parameter) . . . . . . . . . . . . . . . . . . .
Value 2 display (Parameter) . . . . . . . . . . . . . . . . . . .
Value 3 display (Parameter) . . . . . . . . . . . . . . . . . . .
Value 4 display (Parameter) . . . . . . . . . . . . . . . . . . .
Value pressure simulation (Parameter) . . . . . . . . . . .

83
83
84
84
45

W

Web server functionality (Parameter) . . . . . . . . . . . . 76
WizardCommissioning . . . . . . . . . . . . . . . . . . . . . . . . . . 23
Device identification . . . . . . . . . . . . . . 23, 25, 26, 27
Device information . . . . . . . . . . . . . . . . . . . . . . . . 93
Finish . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 95
Firmware update . . . . . . . . . . . . . . . . . . . . . . . . . 92
Measurement adjustments . . . . . . . . . . . . 29, 30, 32
Output settings . . . . . . . . . . . . . . . . . . 33, 34, 35, 37
Performing verification . . . . . . . . . . . . . . . . . . . . . 95
Select file . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 94
Start update . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 92

Z

Zero adjustment (Parameter) . . . . . . . . . . . . . . . . 32, 58
Zero adjustment offset (Parameter) . . . . . . . . . . . . . . 58

T

Target group . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
TCP connection requests (Parameter) . . . . . . . . . . . . . 80
TCP connection timeouts (Parameter) . . . . . . . . . . . . 80
TCP sent packet number (Parameter) . . . . . . . . . . . . . 81
Temperature unit (Parameter) . . . . . . . . . . . . 30, 32, 55
Terminal voltage 1 (Parameter) . . . . . . . . . . . . . . . . . 54
Time zone (Parameter) . . . . . . . . . . . . . . . . . . . . 26, 85
Timestamp (Parameter) . . . . . . . . . . . . . . . . . . . . 38, 39
98

Endress+Hauser

*71606665*
71606665

www.addresses.endress.com