PT2-1583

ガス検知部
７０Ｄシリーズ Ethernet 仕様
通信機能説明書〒174-8744 東京都板橋区小豆沢 2-7-6
ホームページ http://www.rikenkeiki.co.jp/

目次
第１章 スタートアップ ·············································································································································

1

1
1
1
1-2 機器設定 ······························································································································································ 2
1-2-1
GD-70D-EA の IP アドレス設定 ······························································································ 2
1-2-2
パソコンの IP アドレスを設定 ··································································································· 3
1-2-3
通信テスト ········································································································································ 4
1-3 WEB 機能 ····························································································································································· 5
1-3-1
ブラウザの設定······························································································································ 5
1-3-2
ユーザーモードログイン ············································································································· 6
1-3-3
ユーザー管理者モードログイン ······························································································· 7
1-4 PLC 通信機能（OMRON CJ・CS シリーズ） ···························································································· 8
1-4-1
PLC との接続·································································································································· 9
1-4-2
WEB 機能による PLC との通信設定 ····················································································· 10
1-4-3
GD-70D-EA 本体操作による PLC との通信設定 ··························································· 11
1-5 PLC 通信機能（MELSEC Q シリーズ） ····································································································· 13
1-5-1
PLC との接続·································································································································· 14
1-5-2
WEB 機能による PLC との通信設定 ····················································································· 15
1-5-3
GD-70D-EA 本体操作による PLC との通信設定 ··························································· 16
1-1 機器接続 ···································································································································································
1-1-1
PoE スイッチングハブを使用する場合 ·················································································
1-1-2
スイッチングハブ（非 PoE）を使用する場合········································································

第２章 ＷＥＢ機能 ···················································································································································· 18
2-1 使用環境のご注意点 ······································································································································· 18
2-2 機能一覧 ······························································································································································ 20
2-3 ＷＥＢ画面構成··················································································································································· 21
2-3-1
ユーザー画面（Ｕｓｅｒ） ················································································································· 21
2-3-2
ユーザー管理者画面（Authorized Users） ·········································································· 24
2-4 メール送信機能（ガス警報／故障警報） ··································································································· 36.

第３章 Modbus/TCP 通信機能 ·························································································································· 38
3-1 通信仕様 ······························································································································································ 38
3-2 レジスタマップ ····················································································································································· 38
3-3 コマンド ·································································································································································· 43
3-4 例外レスポンス··················································································································································· 44
3-4-1
非対応ファンクションが指定された場合 ·············································································· 44
3-4-2
範囲外のアドレスが指定された場合 ···················································································· 44
3-4-3
範囲外のアドレスへのアクセスが指定された場合 ························································· 44
3-4-4
書き込み不可のアドレスへの書き込みが指定された場合 ·········································· 44

第４章 PLC 通信機能 ············································································································································ 45
4-1 通信仕様 ······························································································································································ 45
4-2 データタイプ ························································································································································· 46
4-2-1
Basic Data：Small ·························································································································· 46
4-2-2
Basic Data：Large ·························································································································· 46
4-2-3
Basic Data：Small+Optional Data ···························································································· 47
4-2-4
Basic Data：Large+Optional Data···························································································· 48
4-3 通信設定内容 ····················································································································································· 49
4-4 データマップ························································································································································· 50
4-4-1
Basic Data：Small ·························································································································· 50
4-4-2
Basic Data：Large ·························································································································· 51
4-4-3
Optional Data ·································································································································· 53
4-5 コマンド ·································································································································································· 54
4-5-1
コマンド 1（設定変更）の発行 ··································································································· 54
4-5-2
コマンド 2～9 の発行（Basic Data：Large のみ） ······························································· 55
4-5-3
コマンド 10（警報テスト濃度設定）の発行（Basic Data：Large のみ） ······················· 56
4-5-4
Basic Data：Large コマンド関連データ ·················································································· 57
4-5-5
Optional Data コマンド関連データ ·························································································· 59
4-5-6
警報点設定変更 ···························································································································· 60
4-5-7
インヒビットの操作 ························································································································ 61
4-6

GD-70D-EA 本体操作による通信設定仕様························································································· 62

1-2．機器設定
1-2-1．GD-70D-EA の IP アドレス設定
GD-70D-EA の電源を ON にします。
ここでは、IP アドレス 192.168.1.1、サブネットマスク 255.255.255.0 に正面パネルのキーより設定
します。
②

キーの説明
① MODE キー①② ▲▼キー③③ SET キー

設定手順
① MODE キー長押しにより、メンテナンスモードにします。
（表示は [1-1／ZERO] ）② ▲▼キーにより、表示を [1-9／M MODE] にし、SET キーを押します。
（表示は [－－－－／M MODE] ）③ 再度 SET キーを長押ししてください。
（表示は [2-0／GAS TEST] ）④ ▲▼キーにより、表示を [2-10／SETTING2 ] にし、SET キーを押します。
（表示は [SET 0／ADDRESS] ）⑤ ▲▼キーにより、表示を [SET 18／ETHERNET ] にし、SET キーを押します。
（表示は [XX／MAC1] ）⑥ ▲▼キーにより、表示を [XXX／IP1] にし、SET キーを押します。
（表示は [XXX／IP1] XXX(数値)が点滅 ）⑦ ▲▼キーにより、IP アドレス表示を [192／IP1] にし、SET キーを押します。
（表示は [192／IP1] ）⑧ ⑥⑦同様に、IP2～4／SUB1～4、をそれぞれ、168、1、1／255、255、255、0 に設定します。
⑨ ▲▼キーにより、表示を [（表示は [／SAVE／SAVE] にし、SET キーを押します。

Y/N] ）⑩ SET キーを押してください。IP アドレスの書き込みを行います。
（IP アドレスの書き込みには 10 秒程度の時間がかかります。）（表示は [SET18／ETHERNET] ）⑪ MODE キー長押しにより、メンテナンスモードを終了します。

2

1-2-2．パソコンの IP アドレスを設定
インターネットプロトコル(TCP/IP)のプロパティで IP アドレスを設定します。
ここでは、IP アドレス 192.168.1.2、サブネットマスク 255.255.255.0 に設定します。
（下図は WindowsXP の例）

注意
通信機器を取り換える際、交換前と交換後の機器の IP アドレスが同一だと、交換してからしばらくの間（通常は 10 分以内）、PC からその機器と通信できない場合があります。
時間を待たずすぐ通信できるようにしたい場合は、「ネットワーク接続」から「修復」を実行します（WindowsXP の場合）。Windows7 の場合は、「修復」が無いので、「ネットワーク接続」から「無効にする」を実行して無効にした後、「有効にする」を実行して元に戻します。

3

1-3-3．ユーザー管理者モードログイン
インターネットエクスプローラーの[アドレス]に次のように入力し、[移動]をクリックしてください。
http：//192.168.1.1/login.html
次の様な画面が表示されるのでデフォルトのパスワード ” GD-70D ” を入力し[Ｌｏｇｉｎ]ボタンをクリックしてください。

ユーザー管理者モードでは、ガス検知器の詳細情報を見ることができ、警報点などの設定変更が
可能です。

7

1-4．PLC 通信機能（OMRON CJ・CS シリーズ）

GD-70D-EA の PLC 通信設定をします。
設定には次の方法があります。
●少点数をフレキシブルに設定したい場合⇒ユーザー管理者ＷＥＢ画面より設定
ＷＥＢ機能の Network 画面から複数の項目をフレキシブルに設定可能です。
[PLC MODE] = 1 になっていない場合、全ての項目を設定変更することはできません。
設定方法の詳細は、1-4-2．ＷＥＢ機能による PLC との通信設定 を参照してください。
●多点数をガス検知器操作で短時間に設定したい場合⇒GD-70D-EA 正面パネルより設定
正面パネルのキーより [PLC MODE] [PLC AREA]の２項目で簡易設定可能です。
PLC の IP アドレス XXX. XXX. XXX.251、NODE251 は固定となります。
（[xxx]は GD-70D-EA の IP アドレスと同じ）設定方法の詳細は、1-4-3．GD-70D-EA 本体操作による PLC との通信設定 を参照してください。

8

1-4-3．GD-70D-EA 本体操作による PLC との通信設定

OMRON CJ・CS シリーズ

GD-70D-EA の電源を ON にします。
[PLC MODE] = 2 [PLC AREA ] =2 に正面パネルのキーより設定します。
PLC（IP アドレス 192.168.1.251、NODE251、FINS/UDP ポート 9600 に設定されているもの）のE2 メモリの 0 アドレスにベーシックデータ（スモール）を書き込む場合の設定
PLC の IP アドレス XXX. XXX. XXX.251、NODE251 は固定となります。
（[xxx]は GD-70D-EA の IP アドレスと同じ）
②
キーの説明
①

① MODE キー③② ▲▼キー③ SET キー

設定手順
① MODE キー長押しにより、メンテナンスモードにします。
（表示は [1-1／ZERO] ）② ▲▼キーにより、表示を [1-9／M MODE] にし、SET キーを押します。
（表示は [－－－－／M MODE] ）③ 再度 SET キーを長押ししてください。
（表示は [2-0／GAS TEST] ）④ ▲▼キーにより、表示を [2-10／SETTING2 ] にし、SET キーを押します。
（表示は [SET 0／ADDRESS] ）⑤ ▲▼キーにより、表示を [SET 18／ETHERNET ] にし、SET キーを押します。
（表示は [XX／MAC1] ）⑥ ▲▼キーにより、表示を [X／PLC MODE] にし、SET キーを押します。
（表示は [X／PLC MODE] X(数値)が点滅 ）⑦ ▲▼キーにより、PLC MODE 表示を [2／PLC MODE] にし、SET キーを押します。
（表示は [2／PLC MODE] ）⑧ ▲▼キーにより、表示を [XX／PLC AREA] にし、SET キーを押します。
（表示は [XX／PLC AREA] XX(数値)が点滅 ）⑨ ▲▼キーにより、PLC AREA 表示を [2／PLC AREA] にし、SET キーを押します（表示は [2／PLC AREA] ）⑩ ▲▼キーにより、表示を [（表示は [／SAVE／SAVE] にし、SET キーを押します。

Y/N] ）⑪ SET キーを押してください。書き込みを行います。
（書き込みには 10 秒程度の時間がかかります。）（表示は [SET18／ETHERNET] ）⑫ MODE キー長押しにより、メンテナンスモードを終了します。

11

OMRON CJ・CS シリーズ設定された項目内容（※GD-70D-EA 本体 IP アドレス 192.168.1.1 の場合）
項目
内容
Type ofMemory Address
PLC MODEPLC Type
Basic Data
Detector write Detector read
2
FINS
Small
0 ※
Area
PLC AREA
Type(FINS)
2
E2[PLC MODE] [PLC AREA] 項目の設定仕様は、4-6．GD-70D-EA 本体操作による通信設定仕様を参照してください。
設定内容の詳細は、第４章 PLC 通信機能 を参照してください。
下図は E2 メモリにデータが書き込まれている様子（E2 の生存カウンタがインクリメントされます。）（CX-Programmer PLC メモリ画面）
PLC の設定、ツールの使い方については、PLC の取扱説明書を参照願います。
注意
PLC のメモリを書き換えるため、間違った設定を行った場合、PLC が想定外の動作をする可能性があります。設定を確定する前に間違いのないことを確認してください。

12

1-5．PLC 通信機能（MELSEC Q シリーズ）

GD-70D-EA の PLC 通信設定をします。
設定には次の方法があります。
●少点数をフレキシブルに設定したい場合⇒ユーザー管理者ＷＥＢ画面より設定
ＷＥＢ機能の Network 画面から複数の項目をフレキシブルに設定可能です。
[PLC MODE] = 1 になっていない場合、全ての項目を設定変更することはできません。
設定方法の詳細は、1-5-2．ＷＥＢ機能による PLC との通信設定 を参照してください。
●多点数をガス検知器操作で短時間に設定したい場合⇒GD-70D-EA 正面パネルより設定
正面パネルのキーより [PLC MODE] [PLC AREA]の２項目で簡易設定可能です。
PLC の IP アドレス XXX. XXX. XXX.251 は固定となります。
（[xxx]は GD-70D-EA の IP アドレスと同じ）設定方法の詳細は、1-5-3．GD-70D-EA 本体操作による PLC との通信設定 を参照してく
ださい。

13

MELSEC Q シリーズ

1-5-3．GD-70D-EA 本体操作による PLC との通信設定
GD-70D-EA の電源を ON にします。

[PLC MODE] = 4 [PLC AREA ] =2 に正面パネルのキーより設定します。
PLC（IP アドレス 192.168.1.251、UDP 自局ポート番号 2000 に設定されているもの）の、ZR メモリの 65536 アドレスにベーシックデータ（スモール）を書き込む場合の設定
PLC の IP アドレス XXX. XXX. XXX.251、NODE251 は固定となります。
（[xxx]は GD-70D-EA の IP アドレスと同じ）
②
キーの説明
①

① MODE キー③② ▲▼キー③ SET キー

設定手順
① MODE キー長押しにより、メンテナンスモードにします。
（表示は [1-1／ZERO] ）② ▲▼キーにより、表示を [1-9／M MODE] にし、SET キーを押します。
（表示は [－－－－／M MODE] ）③ 再度 SET キーを長押ししてください。
（表示は [2-0／GAS TEST] ）④ ▲▼キーにより、表示を [2-10／SETTING2 ] にし、SET キーを押します。
（表示は [SET 0／ADDRESS] ）⑤ ▲▼キーにより、表示を [SET 18／ETHERNET ] にし、SET キーを押します。
（表示は [XX／MAC1] ）⑥ ▲▼キーにより、表示を [X／PLC MODE] にし、SET キーを押します。
（表示は [X／PLC MODE] X(数値)が点滅 ）⑦ ▲▼キーにより、PLC MODE 表示を [4／PLC MODE] にし、SET キーを押します。
（表示は [4／PLC MODE] ）⑧ ▲▼キーにより、表示を [XX／PLC AREA] にし、SET キーを押します。
（表示は [XX／PLC AREA] XX(数値)が点滅 ）⑨ ▲▼キーにより、PLC AREA 表示を [2／PLC AREA] にし、SET キーを押します（表示は [2／PLC AREA] ）⑩ ▲▼キーにより、表示を [（表示は [／SAVE／SAVE] にし、SET キーを押します。

Y/N] ）⑪ SET キーを押してください。書き込みを行います。
（書き込みには 10 秒程度の時間がかかります。）（表示は [SET18／ETHERNET] ）⑫ MODE キー長押しにより、メンテナンスモードを終了します。

16

MELSEC Q シリーズ設定された項目内容（※GD-70D-EA 本体 IP アドレス 192.168.1.1 の場合）
項目
内容
Type ofMemory Address
PLC MODEPLC Type
Basic Data
Detector write Detector read
4
MC
Small
0 ※
Area
PLC AREA
Type(MC)2
ZR(65536)[PLC MODE] [PLC AREA] 項目の設定仕様は、4-6．GD-70D-EA 本体操作による通信設定仕様を参照してください。
設定内容の詳細は、第４章 PLC 通信機能 を参照してください。
下図は ZR メモリにデータが書き込まれている様子（ZR65538 の生存カウンタがインクリメントされます。）（GX－Developer メモリ デバイス一括画面）
PLC の設定、ツールの使い方については、ＰＬＣの取扱説明書を参照願います。
注意
PLC のメモリを書き換えるため、間違った設定を行った場合、PLC が想定外の動作をする可能性があります。設定を確定する前に間違いのないことを確認してください。

17

2-2．機能一覧○：表示

×：表示しない
HOME
ユーザー画面HOMEAUTHORIZED

ユーザー
管理者画面

○

○

Event History

○

○

×

○

Logout×○

Authorized Users の Logout メニュー

Zero Calibration

×

○

ゼロ校正

Span Calibration

×

○

スパン校正

Alarm Test

×○

警報テスト機能

Fault Test

×○History

TEST

機能詳細

Users

StatusCalibration

CALIBREATION

Authorized

各種 GD-70D-EA の設定情報を表示
イベント履歴：100 件のリスト表示，トレンドグラフの表示校正履歴：100 件のリスト表示

故障テスト機能
警報関連の設定内容の表示と設定

Alarm

×

○

Network×○

Date/Time×○

Configuration

×○

MAINTENANCE

×○

メンテナンスモードにする

INHIBIT×○

インヒビット状態にする

ALARM RESET

×○

警報をリセットする

USERS

20

通信／メール／ネットワーク関連の設定内
容の表示と設定
日時の表示と設定
GD-70D-EA 詳細情報及びセンサ情報関連の設定内容の表示と設定

2-3．ＷＥＢ画面構成

Microsoft Internet Explorer を起動しアドレス指定します。
（[xxx]は GD-70D-EA の IP アドレス）
ユーザー画面（Ｕｓｅｒ）
http：//xxx.xxx.xxx.xxx

パスワードなし

ユーザー管理者画面（Authorized Users）
http：//xxx.xxx.xxx.xxx/login.html

パスワードあり

パスワード設定： デフォルトのパスワード “ GD-70D ”
2-3-1．ユーザー画面（Ｕｓｅｒ）
HOME
Status 画面 （Ｕｓｅｒ）

GD-70D-EA の警報や状態をリアルタイムに表示します。
警報設定情報、およびセンサ情報を表示します。
[Open Print Page]ボタンをクリックすると表示しているページの印刷用画面を表示します。

21Event History 画面 （Ｕｓｅｒ）
イベント履歴イベント履歴（警報履歴）を表形式で１画面に 10 件（画面選択で最大 100 件）表示します。
イベント発生時間、イベント内容、トレンドグラフがあるデータは[Graph]ボタンが表示されます。
[Graph]ボタンをクリックすると警報発生の前後、各３分のトレンドグラフを表示します。
[Open Print All]ボタンをクリックすると表示していないページも含め、全イベントの印刷用画面を表示します。
[Open Print Page]ボタンをクリックすると表示しているページの印刷用画面を表示します。
印刷用画面 [Open Print All]

[Open Print Page]

22

Event History 画面 （Ｕｓｅｒ）
トレンドグラフイベント履歴（警報履歴）の[Graph]ボタンをクリックすると警報発生の前後、各３分のトレンドグラ
フを表示します。
[Open Print Page]ボタンをクリックすると表示しているページの印刷用画面を表示します。
[Return]ボタンをクリックするとイベント履歴に戻ります。

23

2-3-2．ユーザー管理者画面（Authorized Users）
HOME
Status 画面 （Authorized User）

GD-70D-EA の警報や状態をリアルタイムに表示します。
警報設定情報、およびセンサ情報を表示します。
警報状態は、[ALARM RESET]ボタンをクリックすることによってリセットすることができます。
[Open Print Page]ボタンをクリックすると表示しているページの印刷用画面を表示します。

24Event History 画面 （Authorized User）
イベント履歴イベント履歴（警報履歴）を表形式で１画面に 10 件（画面選択で最大 100 件）表示します。
イベント発生時間、イベント内容、トレンドグラフがあるデータは[Graph]ボタンが表示されます。
[Graph]ボタンをクリックすると警報発生の前後、各３分のトレンドグラフを表示します。
[Open Print All]ボタンをクリックすると表示していないページも含め、全イベントの印刷用画面を表示します。
[Open Print Page]ボタンをクリックすると表示しているページの印刷用画面を表示します。
印刷用画面 [Open Print All]

[Open Print Page]

25

Event History 画面 （Authorized User）
トレンドグラフイベント履歴（警報履歴）の[Graph]ボタンをクリックすると警報発生の前後、各３分のトレンドグラフ
を表示します。
[Open Print Page]ボタンをクリックすると表示しているページの印刷用画面を表示します。
[Return]ボタンをクリックするとイベント履歴に戻ります。

26

Calibration History 画面 （Authorized User）

校正履歴を表形式で１画面に 10 件（画面選択で最大 100 件）表示します。
[Open Print All]ボタンをクリックすると表示していないページも含め、全履歴の印刷用画面を表示します。
[Open Print Page]ボタンをクリックすると表示しているページの印刷用画面を表示します。

27

CALIBRATION
Zero Calibration 画面 （Authorized User）

ゼロ校正を行うための画面です。
注意
ガス校正については、GD-70Dシリーズ取扱説明をお読みいただき、適切な手段にて行うようお願
いいたします。
弊社では、ガス感度校正などを含めた定期点検、調整、整備等に関するサービスを行っております。試験用標準ガスを作製するには、所定濃度のガスボンベや、ガス袋など専用器具が必要となります。弊社指定のサービス員は、作業を行う上での専用器具や、その他製品に関する専門知識などを備えたスタッフで構成されております。機器の安全動作を維持するために、弊社メンテナン
スサービスをご利用頂きますようお願いします。

28

Span Calibration 画面 （Authorized User）

スパン校正を行うための画面です。
注意
ガス校正については、GD-70Dシリーズ取扱説明をお読みいただき、適切な手段にて行うようお願
いいたします。
弊社では、ガス感度校正などを含めた定期点検、調整、整備等に関するサービスを行っております。試験用標準ガスを作製するには、所定濃度のガスボンベや、ガス袋など専用器具が必要となります。弊社指定のサービス員は、作業を行う上での専用器具や、その他製品に関する専門知識などを備えたスタッフで構成されております。機器の安全動作を維持するために、弊社メンテナン
スサービスをご利用頂きますようお願いします。

29

TEST
Alarm Test 画面 （Authorized User）

警報テストを行うことができます。
注意
警報テストを行うと、GD-70D-EA のガス警報接点出力が動作します。
（警報テスト時、接点動作オンの場合）

30

Fault Test 画面 （Authorized User）

故障テストを行うことができます。
注意
故障テストを行うと、GD-70D-EA の故障警報接点出力が動作します。

31

AUTHORIZED USERS
Alarm 画面 （Authorized User）

警報関連の設定内容の表示と各種設定を行うことができます。
[Update]ボタンをクリックすることによって入力した内容で GD-70D-EA の設定を更新します。

32

Date／Time 画面 （Authorized User）

日時関連の表示と各種設定を行うことができます。

34

Configuration 画面 （Authorized User）

GD-70D-EA のタグ番号、装置名称、設置場所など設定内容の表示と設定を行うことができます。
[Update]ボタンをクリックすることによって入力した内容で検知部の設定を更新します。
タグ番号は、メール機能を使用する場合に、メールの差出人として利用されます。

35

2-4．メール送信機能（ガス警報／故障警報）
AUTHORIZED USERS
Network 画面 （Authorized User）

警報／故障のイベントに対し、GD-70D-EA よりメールアドレス登録先にイベント内容をメール通知
します。
メールの送信元アドレスを１件登録して下さい。メールの送信先は、最大１０件です。送信は登録してある全てのアドレスに対して同時に行います。

36

送信メール(例)
第一警報発報流量異常警報37

第３章 Modbus/TCP 通信機能
3-1．通信仕様
プロトコル

Modbus/TCP

伝送モード

RTU

ポート番号

502番ポート

対応ファンクション

Read Holding Register (0x03)
Preset Multiple Registers (0x10)

同時接続数

８

Modbus/TCP の一般的な仕様に関しては本書の対象外になります。Modbus/TCP の仕様書を参照してください。(http://www.modbus.org/)

3-2．レジスタマップ
GD-70D-EA Modbus/TCP のレジスタマップを以下に示します。
書込
アドレス

○： 可

項目

説明

×：不可
bit0-3：モード（0：イニシャル中

1：測定モード

3：インヒビット

5：テストモード）

bit5：故障フラグ
bit6：アラーム：1st フラグ
40001

×

ステータス

bit7：アラーム：2nd フラグ
bit8：アラーム：1st 接点フラグ
bit9：アラーム：2nd 接点フラグ
bit10：故障接点フラグ
bit11：2 秒周期で 0／1 を繰り返す

40002

×

40003

×

(予約)
浮動小数点
濃度値

40004

×

40005

×

濃度値

40006

×

(予約)

40007

×

濃度単位

アドレス 40003：下位 16 ビット
アドレス 40004：上位 16 ビット
符号付き整数濃度値の小数点以下を四捨五入して整数にした値
bit0-2：小数点コード（000：1/1

001：1/10

010：1/100

011：1/1000）

bit8-11：濃度単位（0001：ppm

0010：ppb

0100：vol%1000：%LEL）

符号無し整数
GD-70D-EA の動作環境(0～40℃)範囲内の値の場合は、温度を示す。範囲外の
40008

×

温度

場合は、40℃より高い、または 0℃未満であることを示す。
・0～40℃：温度(0～40℃)
・41～3276：40℃より高い
・その他：0℃未満

40009

×

(予約)

40010

×

ハートビート

40011

×

(予約)

40012

×

(予約)

40013

×

浮動小数点
警報点 1

40014

×

bit0：ハートビート(1 秒周期で 0/1 を繰り返す)

アドレス 40013：下位 16 ビット
アドレス 40014：上位 16 ビット

38

書込
アドレス

○： 可

項目

説明

×：不可
40015

浮動小数点

×
警報点 2

40016

×

40017

×

警報ステータス40018

×

故障ステータス

40019

×

(予約)

40020

×

(予約)

40021

○

(予約)

40022

○

(予約)

アドレス 40015：下位 16 ビット
アドレス 40016：上位 16 ビット
bit0：アラーム：1st
bit1：アラーム：2nd
bit1：故障

bit15：メンテナンス
bit14：テスト
bit13：インヒビット
bit12：イニシャル
bit11：アラーム：RANGE OVER
bit10：煙検知
40023

×

ステータス

bit9：アラーム：2nd
bit8：アラーム：1st
bit7：異常：センサ
bit6：異常：通信
bit5：異常：流量bit4：注意：流量bit2-3：濃度単位コード（00：vol%bit0-1：小数点コード（00：1/1

01：%LEL

01：1/10

10：ppm10：1/100

11：ppb）
11：1/1000）

符号付き整数
40024

×

濃度値

濃度値の有効数字を整数化したもの。
ステータスの小数点コードの内容をかけた値が実際の濃度値。

40026

×

40025

×

(予約)
(予約)

40027

○

年・月

上位バイト：年（下２桁）

40028

○

日・時

上位バイト：日

下位バイト：時

40029

○

分・秒

上位バイト：分下位バイト：秒

40030

×ハートビート

40031

×

(予約)

40032

×

(予約)

40033

×

温度

40034

×

(予約)

40035

×

(予約)

40036

×

(予約)

40037

×

(予約)

40038

×

(予約)

40039

×

(予約)

40040

×

寿命予測用割合

40041

×

フルスケール

下位バイト：月

bit0：ハートビート(1 秒周期で 0/1 を繰り返す)（アドレス 40010 と同一）符号付き整数
温度を 10 倍した値(単位：℃)

出荷調整時を 100%とした変化割合。
0～100(単位：%)
符号付き整数フルスケールの有効数字を整数化したもの。
ステータスの小数点コードの内容をかけた値が実際のフルスケール。
符号付き整数

40042

×

デジット

デジットの有効数字を整数化したもの。

40043

×

倍率

0：等倍、1：10 分の 1、2：100 分の 1、3：1000 分の 1

40044

×

単位

0：vol%、1：%LEL、2：ppm、3：ppb

ステータスの小数点コードの内容をかけた値が実際のデジット。

39

書込
アドレス

○： 可

項目

説明

×：不可
符号付き整数
40045

○

警報点 1

警報点 1 の有効数字を整数化したもの。
ステータスの小数点コードの内容をかけた値が実際の警報点 1
符号付き整数

40046

○

警報点 2

警報点 2 の有効数字を整数化したもの。

40047

○

(予約)

40048

○

警報遅延時間

単位：10 ミリ秒

40049

◯

故障保持

0：自動復帰

40051

○

警報方式

0：H-HH

40052

○

警報保持

0：自動復帰40053○

テスト時接点動作

0：OFF

40054

○

(予約)

40055

○

励磁/非励磁

40056

◯

(予約)

40057

◯

(予約)

40058

◯

(予約)

40059

○

(予約)

40060

○

(予約)

40061

○

(予約)

40062

×

40063

×

40064

×

40065

×

40066

×

40067

×

40068

×

40069

×

40070

×

40071

×

40072

×

40073

×

40074

×

40075

×

40076

×

40077

×

40078

×

40079

×

40080

×

40081

×

40082

×

40083

×

40084

○

40085

○

40086

○

40087

○

40088

○

40089

○

40090

○

40091

○

40092

○

40093

○

ステータスの小数点コードの内容をかけた値が実際の警報点 2

1：自己保持1：L-LL

2：L-H

1：自己保持

1：ON

bit0：アラーム：1st（0：非励磁、1：励磁）
bit1：アラーム：2nd（0：非励磁、1：励磁）
bit2：故障（0：非励磁、1：励磁）(予約)

シリアル番号

ガス名

タグ番号

ASCII 文字列
前詰め、空白はスペース（0x20）

ASCII 文字列
前詰め、空白はスペース（0x20）

ASCII 文字列
前詰め、空白はスペース（0x20）

40

書込
アドレス

○： 可

項目

説明

×：不可
40094

○

40095

○

40096

○

40097

○

40098

○

40099

○

40100

○

40101

○

40102

○

40103

○

40104

○

40105
40107

○
○
○

40108

○

40109

○

40110

○

40111

○

40112

○

40113

○

40114

○

40115

○

40116

○

40117

○

40118

○

40119

×

40120

×

40121

×

40122

×

40123

×

40124

×

40125

×

40126

×

40127

×

40128

×

40129

×

40130

×

40131

×

40132

×

40133

×

40134

×

40135

×

40136

×

40137

×

40138

×

40139

×

40140

×

40141

×

40142

×

40143

×

40106

装置名称

測定場所

顧客コード

センサシリアル番号

センサ型式

ユニット種別ASCII 文字列
前詰め、空白はスペース（0x20）

ASCII 文字列
前詰め、空白はスペース（0x20）

ASCII 文字列
前詰め、空白はスペース（0x20）

ASCII 文字列
前詰め、空白はスペース（0x20）

ASCII 文字列
前詰め、空白はスペース（0x20）

1：ESU

2：SGU

6：OSU（25%）

3：NCU

7：OSU（5%）(予約)

警報・故障新規フラグ

bit0：アラーム新規フラグ
bit1：故障フラグ

41

4：SSU

5：SSU（煙）8：OSU（50%）

9：SCU

書込
アドレス

○： 可

項目

説明

×：不可
bit0：E-1（センサユニット異常）
bit1-3：予約
bit4：E-5（流量異常）
bit5：E-6（検知器内部通信異常）
bit6：E-7（PLU 異常）

40144

×

故障フラグ

bit7：予約
bit8：E-9（システム異常・時計異常）
bit9：流量注意報（FLOW）
bit10：時計異常（bit8 も同時に ON）
bit11-14：予約
bit15：濃度表示フラグ（GD-70D-EA の LCD 表示が濃度表示とエラーコード表示（E-9 など）を交互に繰り返し表示している状態のとき ON）

40145
～

アドレス 40145 の上位バイト＝第 1 オクテット
×

MAC アドレス

40147
40148
40149
40150
40151
40152
40153
40154

～
アドレス 40147 の下位バイト＝第 6 オクテット
アドレス 40148 の上位バイト＝第 1 オクテット○

IP アドレス～
アドレス 40149 の下位バイト＝第 4 オクテット
アドレス 40150 の上位バイト＝第 1 オクテット◯

サブネットマスク～
アドレス 40151 の下位バイト＝第 4 オクテット◯◯

デフォルトゲートウェ
イ
DHCP

40155

×

寿命判定フラグ

40156

×

使用期限切れフラグ◯（予約）アドレス 40152 の上位バイト＝第 1 オクテット～
アドレス 40153 の下位バイト＝第 4 オクテット
0：OFF

1：ON

上位バイト：寿命予測対応フラグ（0：寿命予測非対応
下位バイト：寿命判定フラグ（0：寿命期限前
0：使用期限前

1：使用期限切れ(期限：3 年)

1：寿命予測対応）

1：寿命期限切れ）（表示：E-1）（表示：E-8）

40157
～
40250
40251

○

コマンド

40252

○

サブコマンド40253○

パラメータ 140254○

パラメータ 240255○

パラメータ 340256○

パラメータ 43-3.コマンド参照

注意
データの書き込み後、５秒以内に次の書き込みを実行すると、レスポンスが遅くなります。
このことを考慮して、タイムアウトの値を設定するか、連続して書き込みを行う場合は５秒以上の間隔をあけるようにしてください。
上記表の （予約） 部分には書き込みを行った場合、予期しない動作をする可能性があります。
（予約） 部分には、書き込みを行わないでください。
アドレス 40003 及び 40004、40005、40024 の濃度値に対し、メンテナンス中の時は、濃度値が負の値になることがあります。
濃度値により数値処理を行う場合は、負の値も考慮した設計を行ってください。

42

3-3．コマンド

アドレス 40251～40256 のレジスタにコマンドを書き込むことで、以下の機能を実行します。
40251

コマンド
MM
(0x4D4D)

GS(0x4753)RA(0x5241)SB(0x5342)

40252
サブ
コマンド

40253
パラメータ
1

40254
パラメータ
2

40255
パラメータ
3

40256
パラメータ
4

機能

S
(0x0053)

－

－

－

－

メンテナンス
モード開始

E(0x0045)

－

－

－

－

メンテナンス
モード終了

0x0000－－－

インヒビット
オフ

0x0001

－

－

－

－

－

－

－

－

－

－

－

濃度値

－

－

－

－

－

－

－

W
(0x0057)
S(0x0053)E(0x0045)W(0x0057)
W(0x0057)

インヒビット
オン
警報テスト
開始
警報テスト
終了
警報テスト
濃度値適用
警報リセット機能の実行は、コマンド（アドレス 40251）への書き込みがトリガとなります。
コマンド、サブコマンド、パラメータ（必要な場合）を、通常は１回のクエリにより実行します。
コマンドだけ書き込んだ場合には、その時点ですでに書き込まれているサブコマンド、パラメータを使って機能を実行します。
注意
上記以外の誤ったコマンドを書き込んだ場合でも、例外レスポンスが返ることはありません。
コマンドの実行結果が失敗の場合も、例外レスポンスが返ることはありません。
ModbusTCP のレスポンスは、書き込みの成否に対するものになります。コマンド実行処理自体の成否については、コマンド実行後に、ステータスや濃度値を読んだ結果により判断してください。

43

3-4．例外レスポンス

3-4-1．非対応ファンクションが指定された場合
データ例：
クエリ

000000000006010400000001

ファンクションコード＝0x04

レスポンス 000000000003018401

例外レスポンス 01（不正ファンクション）

内容説明
GD-70D-EA の対応ファンクションは、Read Holding Register (0x03)、Preset Multiple Registers (0x10)になります。

3-4-2．範囲外のアドレスが指定された場合
データ例：
クエリ

000000000006010301000001

アドレス＝40257

レスポンス 000000000003018302

例外レスポンス 02（不正データアドレス）

内容説明
GD-70D-EA のアドレスは、40001～40256 の範囲です。

3-4-3．範囲外のアドレスへのアクセスが指定された場合
データ例：
クエリ

000000000006010300FF0002

アドレス 40256 から 2 レジスタ読み出し

レスポンス 000000000003018303

例外レスポンス 03（不正データ）

内容説明
アドレス指定は範囲内で、データ数の指定が、最終アドレスが（40256）を超えた場合

3-4-4．書き込み不可のアドレスへの書き込みが指定された場合
データ例：
クエリ

00000000000D0110002E000306000503E80000
アドレス 40047 から 3 レジスタ書き込み （アドレス 40049 が書き込み不可）

レスポンス 000000000003019003

例外レスポンス 03（不正データ）

内容説明
書き込み指定範囲内に一箇所でも書き込み不可アドレスがある場合、ファンクション全体がエラーになり、書き込み可のアドレスも書き込みを行いません。

44

第４章 PLC 通信機能
4-1．通信仕様

プロトコル

FINS／UDP (OMRON PLC 用)、MC／UDP(MELSEC PLC 用)

対応 PLC

OMRON

データタイプ

Basic Data (Small)

CJ・CS シリーズ、MELSEC Q シリーズ

Basic Data (Largel)
Basic Data (Small) + Optional Data
Basic Data (Large) + Optional Data
PLC 構成（推奨）と GD-70D-EA 最大接続台数
通信設定 Interval：min = 1000msec
PLC 構成（推奨）GD-70D-EA 最大接続台数
データタイプ

CPU 型式

イーサネットユニット型式

データタイプ

Basic Data：Large

Basic Data：Small

Basic Data：Small+Optional Data
Basic Data：Large+Optional DataOMRON CJ1H-CPU6□H

CJ1W-ETN21(CJ1W-EIP21)

32

8

OMRON CJ1G-CPU4□H

CJ1W-ETN21(CJ1W-EIP21)

32

8

OMRON CJ2H-CPU6□

CJ1W-ETN21(CJ1W-EIP21)

64

8

OMRON CJ2H-CPU6□-EIP- (CPU 一体型)

64

32

MELSEC Q□□HCPU

QJ71E71-100

8

8MELSEC Q□□UDHCPU

QJ71E71-100

64

8MELSEC Q□□UDEHCPU- (CPU 一体型)

64

8

通信設定 Interval：min = 3000msec
PLC 構成GD-70D-EA 最大接続台数
データタイプ

CPU 型式

イーサネットユニット型式

データタイプ

Basic Data：Large

Basic Data：Small

Basic Data：Small+Optional Data
Basic Data：Large+Optional DataOMRON CJ1H-CPU6□H

CJ1W-ETN21(CJ1W-EIP21)

64

32

OMRON CJ1G-CPU4□H

CJ1W-ETN21(CJ1W-EIP21)

64

32

OMRON CJ2H-CPU6□

CJ1W-ETN21(CJ1W-EIP21)

128

64

OMRON CJ2H-CPU6□-EIP- (CPU 一体型)

128

64

MELSEC Q□□HCPU

QJ71E71-100

64

32

MELSEC Q□□UDHCPU

QJ71E71-100

128

64

MELSEC Q□□UDEHCPU- (CPU 一体型)

128

64

PLC の取扱・仕様に関しては本書の対象外になります。ご使用になる PLC の取扱説明書を参照してください。

45

4-2．データタイプ

4-2-1 Basic Data：Small
ベーシックデータ：スモール
警報ステータス、濃度値、警報点などの基本的な情報の書き込み

PLC メモリ(D,E など)
GD-70D-EA

GD-70D-EA 書き込み
先頭アドレス

16WORD
Basic Data：Small
GD-70D-EA
から PLC へ書き込み

4-2-2 Basic Data：Large
ベーシックデータ：ラージ
警報ステータス、濃度値、警報点などの基本的な情報と、日時、ガス名などの書き込み
警報動作設定、警報点、などの設定変更のためのデータ読み込み

PLC メモリ(D,E など)
GD-70D-EA

GD-70D-EA 書き込み
先頭アドレス

42WORD
Basic Data：Large
GD-70D-EA
から PLC へ
書き込み

22WORD
Basic Data：Large
GD-70D-EA
が PLC から読み込み

46

GD-70D-EA 読み込み
先頭アドレス

4-2-3 Basic Data：Small + Optional Data
ベーシックデータ：スモール+オプショナルデータ
警報ステータス、濃度値、警報点などの基本的な情報の書き込み
各種名称、センサ型式、センサシリアル番号などの拡張データの読み込み
各種名称などの変更のためのデータ読み込み

PLC メモリ(D,E など)
GD-70D-EA

GD-70D-EA 書き込み
先頭アドレス

16WORD
Basic Data：Small
72WORD
Optional Data

GD-70D-EA
から PLC へ
書き込み

GD-70D-EA 読み込み
先頭アドレス
56WORD
Optional Data

GD-70D-EA
が PLC から
読み込み

47

4-2-4 Basic Data：Large + Optional Data
ベーシックデータ：ラージ+オプショナルデータ
警報ステータス、濃度値、警報点などの基本的な情報と、日時、ガス名などの書き込み
警報動作設定、警報点、などの設定変更のためのデータ読み込み
各種名称、センサ型式、センサシリアル番号などの拡張データの読み込み
各種名称などの変更のためのデータ読み込み

PLC メモリ(D,E など)
GD-70D-EA

GD-70D-EA 書き込み
先頭アドレス

42WORD
Basic Data：Large

72WORD
Optional Data
GD-70D-EA
から PLC へ
書き込み

GD-70D-EA 読み込み
先頭アドレス
22WORD
Basic Data：Large
56WORD
Optional Data
GD-70D-EA
が PLC から読み込み

48

4-3．通信設定内容

GD-70D-EA の WEB 機能よりユーザー管理者でログインした Network 画面又は正面パネルのキーより設定を行います。但し、正面パネルからの設定には制限があります。
WEB 機能の詳細は、第２章 WEB 機能 を参照してください。正面パネルからの通信設定仕様詳細は、
4-6．GD-70D-EA 本体操作による通信設定仕様 を参照してください。
項目
PLC

PLC Type

Type of Basic Data
Use Optional Data

説明

備考

PLC の通信タイプを選択します。

Not Used：使用しない(デフォルト)

FINS を選択した場合は本表の FINS の設定、MC をFINS：OMRON CJ・CS シリーズ

選択した場合は本表の MC の設定が使用されます。

MC：MELSEC Q シリーズ

Small

：スモール

Largel：ラージ

ベーシックデータの種別
オプションデータの使用有無

MinInterval
Max

Yes

：使用する

No

：使用しない（デフォルト）

最少送信間隔。データ変化時、通常測定時でない場

250～10000msec

合（警報時、メンテナンス状態時）の送信間隔。

最小時間のデフォルト

最大送信間隔。通常測定時で、データに変化がない
場合の送信間隔

最大時間のデフォルト

PLC への送信に対して、指定した時間が経過しても1～10sec

応答が無い場合、応答待ちを中止します。Interval で

デフォルト

：5sec

デフォルト

：9600

デフォルト

：5000

設定した時間が経過すると、再度送信します。
FINS

IP Address

PLC の IP アドレス

Port

UDP ポート番号

Detector

Network

GD-70D-EA のネットワークアドレス

Address（FINS 仕様上の設定値）

Node Address
Unit Number
PLC（FINS 仕様上の設定値）（FINS 仕様上の設定値）

MemoryAddress
MemoryAddress
Area TypeMC

GD-70D-EA のユニット番号
PLC のネットワークアドレス

Area Type
Detector Read

（FINS 仕様上の設定値）

Address

Unit Number
Detector Write

GD-70D-EA のノードアドレス

Network

Node Address

PLC のノードアドレス（FINS 仕様上の設定値）
PLC のユニット番号（FINS 仕様上の設定値）
PLC への書き込みメモリアドレス
PLC への書き込みメモリエリア
PLC からの読み込みメモリアドレス
PLC からの読み込みメモリエリア

IP Address

PLC のＩＰアドレスPort

UDP ポート番号

PLC

Network

PLC のネットワークアドレス

Address（MC 仕様上の設定値）

Node Address
Detector Write

MemoryAddress
Area Type

Detector Read

MemoryAddress
Area Type：3000msec（濃度変化無し）

PLC からの受信タイムアウト時間
Timeout：1000msec（濃度変化有り）

PLC のノードアドレス（MC 仕様上の設定値）
PLC への書き込み先頭メモリアドレス
PLC への書き込みメモリエリア
PLC からの読み込み先頭メモリアドレス
PLC からの読み込みメモリエリア

49

4-4．データマップ

4-4-1 Basic Data：Small
書き込みデータマップ
アドレス

項目

説明

GD-70D-EA

bit15：メンテナンス

データ

bit14：テスト
bit13：インヒビット
bit12：イニシャル
bit11：アラーム：RANGE OVER
bit10：煙検知0

bit9：アラーム：2nd

ステータス

bit8：アラーム：1st
bit7：異常：センサ
bit6：異常：通信
bit5：異常：流量bit4：注意：流量bit2-3：濃度単位コード（00：vol%bit0-1：小数点コード（00：1/1

01：%LEL

01：1/10

10：ppm10：1/100

11：ppb）
11：1/1000）

符号付き整数1

濃度値

濃度値の有効数字を整数化したもの。
ステータスの小数点コードの内容をかけた値が実際の濃度値。

GD-70D-EA

生存確認用カウンタ

3

流量

4

熱分解器温度

単位は、℃。熱分解器がない場合は、0x8000。

IP アドレス

IP アドレスの前半。

（上位 16 ビット）例）192.168.0.1 の場合、0xC0A8

IP アドレス

IP アドレスの後半。

（下位 16 ビット）例）192.168.0.1 の場合、0x0001

サブネットマスク

サブネットマスクの前半。

（上位 16 ビット）

255.225.225.0 の場合、0xFFFF

サブネットマスク

サブネットマスクの後半。

（下位 16 ビット）

255.225.225.0 の場合、0xFF00

5

書き込み
6

領域

送信ごとに増加する値。

2

7
8

9

オーバーフロー時は、０にもどる。
単位は、ミリリットル／分。

bit6：感度補正（0：オフ

1：オン）

bit5：ゼロ追尾（0：オフ

1：オン）

bit4：サプレスタイプ（0：Cut off

フラグ

1：Slope）

bit3：故障警報動作（0：自動復帰

1：自己保持）bit2：ガス警報動作（0：自動復帰

1：自己保持）bit0-1：警報タイプ（0：H-HH

1：L-LL

2：L-H）

符号付き整数10

フルスケールステータスの小数点コードの内容をかけた値が実際のフルスケー
ル。
符号付き整数11

1st 警報点

12

2nd 警報点13

濃度値 3200 分割

フルスケールを 3200 とした時の相対値。

14

1st 警報点 3200 分割

フルスケールを 3200 とした時の相対値。

15

2nd 警報点 3200 分割

フルスケールを 3200 とした時の相対値。

ステータスの小数点コードの内容をかけた値が実際の 1st 警報点。
符号付き整数
ステータスの小数点コードの内容をかけた値が実際の 2nd 警報点。

注意アドレス 1 の濃度値に対し、メンテナンス中の時は、濃度値が負の値になることがあります。
濃度値により数値処理を行う場合は、負の値も考慮した設計を行ってください。
50

4-4-2 Basic Data：Large
書き込みデータマップ
アドレス
GD-70D-EA
データ

項目

説明

濃度・警報ステータスな

Basic Data(Small)と同一

ど(4-1-1 Basic Data：Small 参照)

16

日時（年・月）

上位バイト：年（下２桁）

17

日時（日・時）上位バイト：日

下位バイト：時

18

日時（分・秒）

上位バイト：分下位バイト：秒

19

警報遅延時間

0～15

20

デジット

21

ゼロサプレス値

下位バイト：月単位は、10 ミリ秒。
符号付き整数
ステータスの小数点コードの内容をかけた値が実際のデジット。
符号付き整数
ステータスの小数点コードの内容をかけた値が実際のゼロサプ
レス値。

22～27

ガス名

28～34

（予約）

35

（予約）

ASCII 文字列
前詰め、空白はスペース（0x20）

bit0-7：寿命予測割合
bit8-12：予約
36

寿命データ

bit13：使用済み判定
bit14：寿命判定

GD-70D-EA

bit15：寿命機能有無

書き込み
領域

37～39
コマンド
データ

40

（予約）
コマンド実行ステータス（GD-70D-EA）

0： 通常状態
1： 処理中コマンド 1（設定変更）の時(成功時 bit オン 失敗時 bit オフ)
bit12：感度補正
bit11：日時
bit10：ゼロ追尾
bit9：ゼロサプレスタイプ
bit8：ゼロサプレス値
bit7：警報遅延時間
bit6：故障警報動作

41

bit5：ガス警報動作

コマンド実行結果

bit4：2nd 警報点
bit3：1st 警報点
bit2：警報タイプ
bit1：サブネットマスク
bit0：IP アドレス
コマンド 2～
0： 失敗
1： 成功

注意
アドレス 1 の濃度値に対し、メンテナンス中の時は、濃度値が負の値になることがあります。
濃度値により数値処理を行う場合は、負の値も考慮した設計を行ってください。

51

読み込みデータマップ
アドレス

項目

説明

GD-70D-EA
設定
データ

0

1
2
3

IP アドレス

IP アドレスの前半。

（上位 16 ビット）
IP アドレス

IP アドレスの後半。

（下位 16 ビット）
サブネットマスク

サブネットマスクの前半。

（上位 16 ビット）
サブネットマスク

サブネットマスクの後半。

（下位 16 ビット）

bit3：故障警報動作（0：自動復帰、1：自己保持）4

フラグ

bit2：ガス警報動作（0：自動復帰、1：自己保持）
bit0-1：警報タイプ（0：H-HH、1：L-LL、2：L-H）
符号付き整数5

1st 警報点

ステータスの小数点コードの内容をかけた値が実際の 1st 警報
点。
符号付き整数6

GD-70D-EA

2nd 警報点

ステータスの小数点コードの内容をかけた値が実際の 2nd 警報
点。

読み込み
領域

7

日時（年・月）

上位バイト：年（下２桁）

下位バイト：月8

日時（日・時）上位バイト：日

下位バイト：時

9日時（分・秒）

上位バイト：分下位バイト：秒

10

警報遅延時間

単位は、10 ミリ秒。

11

ゼロサプレス値

符号付き整数
ステータスの小数点コードの内容をかけた値が実際のゼロサプ
レス値。
12～15
コマンド
データ

16

（予約）
コマンド実行ステータス

0：通常状態(PLC)

1：処理実行

17

コマンドコード

18

パラメータ

19～21

（予約）

(4-5 コマンド参照)

52

4-4-3 Optional Data
書き込みデータマップ

GD-70D-EA
データ

アドレス

アドレス

Basic Data

Basic Data

(Small)(Large)

の場合

の場合

16～25

42～51

TAG 番号

26～35

52～61

装置名称

36～45

62～71

測定場所

46～50

72～76

顧客コード

51～55

77～81

センサ型式

56～65

82～91

センサシリアル番号

66～67

92～93

ユニット種別設定

68～85

94～111

（予約）

86

112

項目

GD-70D-EA
書き込み
領域
コマンド
データ

説明

ASCII 文字列
前詰め、空白は、スペース（0x20）
ASCII 文字列
前詰め、空白は、スペース（0x20）
ASCII 文字列
前詰め、空白は、スペース（0x20）
ASCII 文字列
前詰め、空白は、スペース（0x20）
ASCII 文字列
前詰め、空白は、スペース（0x20）
ASCII 文字列
前詰め、空白は、スペース（0x20）
ASCII 文字列
前詰め３文字コマンド実行ステータス

0：通常状態（GD-70D-EA）

1：処理完了
コマンド 1（設定変更）の時(成功時 bit オン
失敗時 bit オフ)
bit3：顧客コード
bit2：測定場所

87

113

コマンド実行結果

bit1：装置名称
bit0：TAG 番号
コマンド 2～（予約）
0： 失敗
1： 成功

読み込みデータマップ

GD-70D-EA
設定
データ

アドレス

アドレス

Basic Data

Basic Data

(Small)(Large)

の場合

の場合

0～9

22～31

TAG 番号

10～19

32～41

装置名称

20～29

42～51

測定場所

30～34

52～56

顧客コード

35～49

57～71

（予約）

50

72

51

73

コマンドコード

52

74

パラメータ

53～55

75～77

（予約）

項目

GD-70D-EA
読み込み
領域
コマンド
データ

説明

ASCII 文字列
前詰め、空白は、スペース（0x20）
ASCII 文字列
前詰め、空白は、スペース（0x20）
ASCII 文字列
前詰め、空白は、スペース（0x20）
ASCII 文字列
前詰め、空白は、スペース（0x20）

コマンド実行ステータス

0：通常状態(PLC)

1：処理実行

53(4-5 コマンド参照)

4-5．コマンド

Basic Data：Large および Optional Data の GD-70D-EA 設定データ、コマンド実行ステータス（PLC）、コ
マンドコード、パラメータに書き込むことで、各種の機能を実行します。
4-5-1 コマンド 1（設定変更）の発行[コマンド実行ステータス(PLC)] = 0（通常状態）なことを確認する。
↓[ GD-70D-EA 設定データ ] ← 変更したい項目にデータを設定する。（複数設定可）[

コマンドコード] ← 1（設定変更）を設定する。

[

パラメータ

] ← 変更したい項目のビットをオンにした値を設定する。
（複数変更したい場合は複数ビットをオン）↓[コマンド実行ステータス(PLC)] ← 1（処理実行）を設定する。
↓[コマンド実行ステータス(GD-70D-EA) ] = 1（処理完了）なことを確認する。
↓[コマンド実行ステータス(PLC)] ← 0（通常状態）を設定する。
設定変更に成功した場合 [コマンド実行結果] = 変更項目のビットがオン
設定変更に失敗した場合 [コマンド実行結果] = 変更項目のビットがオフ注意
コマンドを発行してから、処理が完了するまでは、[GD-70D-EA 設定データ] [コマンドコード][パラメータ]の値の変更はおこなわないでください。

54

4-5-2 コマンド 2～9 の発行（Basic Data：Large のみ）[コマンド実行ステータス(PLC)] = 0（通常状態）なことを確認する。
↓[ GD-70D-EA 設定データ ] ← 変更したい項目にデータを設定する。（複数設定可）[

コマンドコード] ← 2～9 を設定する。

[

パラメータ

] ← 0 を設定する。（未使用）（コマンド 2～9）↓

[コマンド実行ステータス(PLC)] ← 1（処理実行）を設定する。
↓[コマンド実行ステータス(GD-70D-EA) ] = 1（処理完了）なことを確認する。
↓[コマンド実行ステータス(PLC)] ← 0（通常状態）を設定する。
設定変更に成功した場合 [コマンド実行結果] = 1
設定変更に失敗した場合 [コマンド実行結果] = 0
注意
コマンドを発行してから、処理が完了するまでは、[GD-70D-EA 設定データ] [コマンドコード][パラメータ]の値の変更はおこなわないでください。

55

4-5-3 コマンド 10（警報テスト濃度設定）の発行（Basic Data：Large のみ）[コマンド実行ステータス(PLC)] = 0（通常状態）なことを確認する。
↓[ GD-70D-EA 設定データ ] ← 変更したい項目にデータを設定する。（複数設定可）[

コマンドコード] ← 10（警報テスト濃度設定）を設定する。

[

パラメータ

] ← 警報テスト濃度値を設定する。
↓[コマンド実行ステータス(PLC)] ← 1（処理実行）を設定する。
↓[コマンド実行ステータス(GD-70D-EA) ] = 1（処理完了）なことを確認する。
↓[コマンド実行ステータス(PLC)] ← 0（通常状態）を設定する。
設定変更に成功した場合 [コマンド実行結果] = 1
設定変更に失敗した場合 [コマンド実行結果] = 0
注意
コマンドを発行してから、処理が完了するまでは、[GD-70D-EA 設定データ] [コマンドコード][パラメータ]の値の変更はおこなわないでください。

56

4-5-4 Basic Data：Large コマンド関連データ
項目

説明

40

コマンド実行ステータス（GD-70D-EA）

41

コマンド実行結果

0： 通常状態
1： 処理中コマンド 1（設定変更）の時(成功時 bit オン
失敗時 bit オフ)
bit12：感度補正
bit11：日時
bit10：ゼロ追尾
bit9：ゼロサプレスタイプ
bit8：ゼロサプレス値
bit7：警報遅延時間
bit6：故障警報動作
bit5：ガス警報動作
bit4：2nd 警報点
bit3：1st 警報点
bit2：警報タイプ
bit1：サブネットマスク
bit0：IP アドレス

アドレス

GD-70D-EA
書き込み
領域

コマンド 2～
0：失敗
1：成功
アドレス
0～15
GD-70D-EA
読み込み
領域

項目

説明

各種設定データ

16

コマンド実行ステータス(PLC)

0：通常状態
1：処理実行

17

コマンド

コマンドコードを設定する。

18

パラメータ

コマンドによる。（下記コマンドコード表参照）

Basic Data：Large のコマンドコード
コマンドコード
1
2
3
4
5
6
7
8
9
10

内容
設定変更
ブザーストップ
インヒビット オン
インヒビットオフ
メンテナンスモード開始
メンテナンスモード終了
ゼロ校正実行
警報テストモード開始
警報テストモード終了
警報テスト濃度設定

パラメータ
設定フラグ（下表参照）
未使用
未使用
未使用
未使用
未使用
未使用
未使用
未使用
テスト濃度値57

コマンドコード１（設定変更）時のパラメータ（設定フラグ）
同時に複数の項目を変更することも可能です。この場合は同時に複数の bit をオンにしてください。
パラメータ（ビット）
bit15
Bit14
Bit13
Bit12
bit11
bit10
bit9
bit8
bit7
bit6
bit5
bit4
Bit3
Bit2
Bit1
Bit0

内容
（予約）
（予約）
（予約）
感度補正
日時
ゼロ追尾
ゼロサプレスタイプ
ゼロサプレス値
警報遅延時間故障警報動作ガス警報動作
2nd 警報点
1st 警報点
警報タイプ
サブネットマスク
IP アドレス58

4-5-5 Optional Data コマンド関連データ

GD-70D-EA
書き込み
領域

GD-70D-EA
読み込み
領域

アドレス
Basic Data
(Small)の場合

アドレス
Basic Data
(Large)
の場合

86

項目

説明

102

コマンド実行ステータス（GD-70D-EA）

87

103

コマンド実行結果

0：通常状態
1：処理中
コマンド 1（設定変更）の時(成功時 bit オン
失敗時 bit オフ)
bit3：顧客コード
bit2：測定場所
bit1：装置名称
bit0：TAG 番号
コマンド 2～（予約）
0：NG
1：OK

アドレス
Basic Data
(Small)の場合

アドレス
Basic Data
(Large)
の場合

0～49

22～71

50

72

コマンド実行ステータ
ス(PLC)

0：通常状態
1：処理中

17

73

コマンド

コマンドコードを設定する。

18

74

パラメータ

コマンドによる。（下記コマンドコード表参照）

説明項目

各種設定データ

Optional Data のコマンドコード
コマンドコード
1

内容
設定変更

パラメータ
設定フラグ（下表参照）

コマンドコード１（設定変更）時のパラメータ（設定フラグ）
パラメータ（ビット）
bit15
Bit14
Bit13
Bit12
bit11
bit10
bit9
bit8
bit7
bit6
bit5
bit4
Bit3
Bit2
Bit1
Bit0

内容
（予約）
（予約）
（予約）
（予約）
（予約）
（予約）
（予約）
（予約）
（予約）
（予約）
（予約）
（予約）
顧客コード
測定場所
装置名称
TAG 番号

59

4-5-6 警報点設定変更
1st 警報点を 500、2nd 警報点を 1000 に設定する場合
アドレス[ GD-70D-EA 設定データ（1st 警報点）]：Basic Data：Large GD-70D-EA 読み込み領域アドレス 5[コマンド実行ステータス(PLC)]：Basic Data：Large GD-70D-EA 読み込み領域アドレス 16[

コマンドコード]：Basic Data：Large GD-70D-EA 読み込み領域アドレス 17

[

パラメータ

]

：Basic Data：Large GD-70D-EA 読み込み領域アドレス 18[コマンド実行ステータス(GD-70D-EA) ]：Basic Data：Large GD-70D-EA 書き込み領域アドレス 40[コマンド実行結果]：Basic Data：Large GD-70D-EA 書き込み領域アドレス 41[コマンド実行ステータス(PLC)] = 0（通常状態）なことを確認する。
↓[ GD-70D-EA 設定データ ] ← 500[

コマンドコード] ← 1（設定変更）を設定する。

[

パラメータ

] ← 0x0018

設定フラグ 1st 警報点(bit3)と 2nd 警報点(bit4)をオンにした値↓[コマンド実行ステータス(PLC)] ← 1（処理実行）を設定する。
↓[コマンド実行ステータス(GD-70D-EA) ] = 1（処理完了）なことを確認する。
↓[コマンド実行ステータス(PLC)] ← 0（通常状態）を設定する。
設定変更に成功した場合 [コマンド実行結果] = 0018
1st 警報点(bit3)と 2nd 警報点(bit4)がオン
設定変更に失敗した場合 [コマンド実行結果] = 0000
1st 警報点(bit3)と 2nd 警報点(bit4)がオフ

60

4-5-7 インヒビットの操作
アドレス[コマンド実行ステータス(PLC)]：Basic Data：Large GD-70D-EA 読み込み領域アドレス 16[

コマンドコード]：Basic Data：Large GD-70D-EA 読み込み領域アドレス 17

[

パラメータ

]

：Basic Data：Large GD-70D-EA 読み込み領域アドレス 18[コマンド実行ステータス(GD-70D-EA) ]：Basic Data：Large GD-70D-EA 書き込み領域アドレス 40[コマンド実行結果]：Basic Data：Large GD-70D-EA 書き込み領域アドレス 41[コマンド実行ステータス(PLC)] = 0（通常状態）なことを確認する。
↓[

コマンドコード] ← 3（インヒビットオン）を設定する。
（インヒビットオフの場合は 4）

[

パラメータ

] ← 0（パラメータ未使用）↓[コマンド実行ステータス(PLC)] ← 1（処理実行）を設定する。
↓[コマンド実行ステータス(GD-70D-EA) ] = 1（処理完了）なことを確認する。
↓[コマンド実行ステータス(PLC)] ← 0（通常状態）を設定する。
設定変更に成功した場合 [コマンド実行結果] = 1
設定変更に失敗した場合 [コマンド実行結果] = 061