PDI

The latest model of Levelnic series. Built-in wireless.
Measurement value can be checked both on PC screen and the device's display panel.

HOW TO MOVIEANGLELEVELNIC DL-S4W SERIESFor vertical measurement▶
L-type Base■ High precision Digital Level■ Resolution 0.001mm/m■ Switchable between wired
and wireless■ Stainless steel base typethat not requires rustprevention maintenance（DL-S4W SUS）

P127
For setting a fixed pitch▶
Levelnic Base with Adjustable Pitch

P137
For flatness andstraightness calculation▶
Levelnic Measurement Software

P138
〜139

Rust prevention▶
Zerust Film for SteelP380

DL-S4W
DL-S4W SUSUSE●High precision leveling of machine tools and surface plates. Straightness and flatness measurement.
MATERIAL●Body：AluminumFEATURES●Wireless data transmission●PC screen becomes display of Levelnic●Measurement data can be displayed on PC, using receiver and data display software●Switchable between wireless / wired for saving of electricity and long-time use●USB receiving unit installed. The driver is supported by the automatic recognition system.
●Equipped with two rechargeable batteries, enables running measuring by using a spare battery while the other is being charged●Custom-made base can be attached to the threaded screw holes（M5 × P0.8）at the bottom base
SPECIFICATIONSPRECISION
LEVEL

Measuring Range※1± 5.00mm/m± 0.286°

Minimum Reading

0.001mm/m

0.0001°（within ± 1.999mm/m）

0.01mm/m

LEVELNIC

0.001°（over ± 2mm/m）

ANGLE

Operating Temperature

0 ～ 40℃

Accuracy※2

Operating temperature 17 ～ 23℃ ± 0.85% rdg （within ± 1.999mm/m）
Operating temperature 0 ～ 40℃

LEVEL± 1.0% rdg （over ± 2mm/m）±
 2.6% rdg （within ± 1.999mm/m）

PROTRACTOR± 2.7% rdg （over ± 2mm/m）

Repeatability

Within ± 0.005mm/m± 0.0003°

Output Signal

Conforms to RS-232C, wired ※4, built-in wireless system（Bluetooth Class1 ※5, wireless range： 30 ～ 50m）

Power Source

9V 006P type dry cell 1pc

8.4V 006P type rechargeable battery 1pc

AC adapter（DC9V output ・AC100V ～ 240V）included
Run time※3（continuous use）

Zinc-carbon cell： Approx. 12 hrs.（wireless use approx. 3 hrs.）
Alkaline cell： Approx. 30 hrs.（wireless use approx. 10 hrs.）
Rechargeable battery 10 hrs.（wireless use approx. 4 hrs.）

Dimensions

L158 × W61 × H107mm（Base dimensions： L150 × W55mm）

Accessories● Shock resistant aluminum case● Warranty・Instruction manual・Inspection report● AC adapter（DC9V output ・AC100V ～ 240V）● Battery charger● Dedicated display software● 8.4V 006P type rechargeable battery 2pcs
● USB receiving unit

※ 1： F or DEG（°）Display mode,“0（Zero）”before the decimal point is not displayed.
For 0.001mm/m（0.0001°）display mode, maximum reading which can be displayed is less than ± 1.999mm/m（± 0.1145°）.
※ 2： % rdg indicates the percentage of displayed value※ 3： May vary depending upon measurement conditions※ 4： PC without D-USB pin port（RS-232C）requires USB serial transmitter（optional）※ 5： Our Bluetooth system qualification is with TELEC（Japan）, FCC（U.S.A）, CE（EU）, KCC（South Korea）
Model No.

010053DL-S4W

010111DL-S4W SUS

010085

DL-S4W-E

Software Language

Japanese
EnglishBase Material

WeightCast iron

2.0kg

Stainless steel SUS303

2.0kg

Cast iron

2.0kg

125

再 校

02_121-160角度.indd 125

CMYK

2019/02/26 13:08

19/02/26 13:08:56

Order No.

ANGLELEVELNIC DL-S3 SERIESPDIHigh accuracy Level with simple & clear operation.
The world's first Digital Level with built-in Microcomputer.

HOW TO MOVIE■ High precision Digital Level■ Resolution 0.001mm/m■ Rechargable Battery（DL-S3C / DL-S3C SUS）
■ Stainless steel base type that notrequires rust prevention maintenance（DL-S3 SUS / DL-S3C SUS）

For vertical measurement▶

For wireless communication▶

L-type Base

Wireless Adapter

P127
For display measured value
for two connections▶
Transmitter and display set

P131
For setting a fixed pitch▶
Levelnic Base with Adjustable Pitch

P132
For flatness andstraightness calculation▶
Levelnic Measurement Software

P137
Rust prevention▶
Zerust Film for Steel

P138
〜139

P380
USE
●High precision leveling of machine tools and surface plates.
Straightness, flatness and vertical measurement.
MATERIAL●Body：AluminumFEATURES●Easy to fix the datum point, divide measurement value into halves with one-touch operation（with "0" cal. switch, 1/2 cal. switch）●Measurement value can be output with RS232 signal●Custom-made base can be attached to the threaded screw holes（M5 × P0.8）at the bottom base●DL-S3C is rechargeable type.  It can be charged when not in use.
SPECIFICATIONS
Model No.

DL-S3/DL-S3 SUS

DL-S3C/DL-S3C SUS

Measuring Range※1± 5.00mm/m・± 0.286°

Minimum Reading

0.001mm/m・0.0001°（0 ～± 2mm/m・0 ～± 0.115°）
0.01mm/m・0.001°

LEVELNICPRECISION
LEVEL
ANGLE

LEVEL
PROTRA
CTOR

Operating Temperature

0 〜 40℃

Accuracy ※2

Operating temperature17 〜 23℃ ± 0.85%rdg（0 ～± 2mm/m・0 ～± 0.115°）± 1.0%rdg（± 2 〜± 5mm/m・± 0.115°〜± 0.286°）
Operating temperature 0 〜 40℃ ± 2.6%rdg（0 〜± 2mm/m・0 〜± 0.115°）± 2.7%rdg（± 2 〜± 5mm/m・± 0.115°〜± 0.286°）

Repeatability

Within ± 0.005mm/m・± 0.0003°

Output Signal

Conforms to RS-232C ※4

Charging time−Power Source

9V 006P type dry cell 1pc

12 hrs.（when fully charged）
Ni-MH battery（8.4V 200mAh）

AC adapter（DC9V output ・AC100V ～ 240V）included
Run time※3（continuous use）

Zinc-carbon cell： Approx. 10 hrs.

Dimensions

L172 × W65 × H129mm（base dimension： L150 × W55mm）

Accessories● Shock resistant aluminum case● Warranty・Instruction manual・Inspection report● AC adapter（DC9V output ・AC100V ～ 240V）● 9V dry cell

12 hrs. charge： Approx. 9 hrs.

Alkaline cell： Approx. 20 hrs.

8 hrs. charge： Approx. 6 hrs.
4 hrs. charge： Approx. 2 hrs.

● Shock resistant aluminum case● Warranty・Instruction manual・Inspection report● AC adapter（DC15V output）● Ni-MH battery（8.4V 200mAh）※ 1： F or DEG（°）Display mode, "0（Zero）" before the decimal point is not displayed.
For 0.001mm/m（0.0001°）display mode, maximum reading which can be displayed is less than ± 1.999mm/m（± 0.1145°）.
※ 2： % rdg indicates the percentage of displayed value※ 3： May vary depending upon measurement conditions※ 4： PC without D-USB pin port（RS-232C）requires USB serial transmitter（optional）

19/02/26 13:08:59

Order No.

Model No.

Battery Type

010013DL-S3

9V dry cell

010030DL-S3C

Ni-MH battery

010112

DL-S3 SUS

9V dry cell

010113DL-S3C SUS

Ni-MH battery

Base Material

Cast iron
Stainless steel SUS303

Weight

2.2kg
2.2kg
2.2kg
2.2kg

With V groove base - available on request
Model

DL-S3V

Tolerance

Within 0.1mm/m（ on condition that the bubble in the sub-bubble tube is within the base line）

Tolerance Between Flat and V Groove

Within 0.05mm/m

Angle of V Groove130°（ for cylindrical objects up to φ 65mm ）

126

再 校

02_121-160角度.indd 126

CMYK

2019/02/26 13:09

DL-S3LPDI

LEVELNIC DL-S3 with magnetic L-type base.
Magnetic base enables Z-axis measurement in addition to X-axis and Y-axis.

HOW TO MOVIEANGLELEVELNIC

For flatness andstraightness calculation▶
Levelnic Measurement Software

P138
〜139

USE
●High precision leveling of machine tools and surface plates.
Straightness, flatness and vertical measurement.
MATERIAL●DL-S3 / Body：Aluminum●L-type base：Cast iron
FEATURES●Quick digital display of the measured value●Measurement value can be output with RS-232 signal●L-type magnetic base enables vertical（Z-axis）measurement●With 130°
V groove
POWER●9V dry cell 1pc●AC adapter（DC9V output・AC100V～240V）
ACCESSORIES●Shock resistant aluminum case
●Warranty・Instruction manual・Inspection report●AC adapter（DC9V output・AC100V～240V）●9V dry cell

Please refer to DL-S3（P126）for detailed specifications for this unit

Order No.

Model No.

Base Squareness（mm）

Weight

010027

DL-S3L

Within ± 0.04

5.5kg

L-TYPE BASE FOR LEVELNICLEVELNICPRECISION
LEVEL

DL-SL

ANGLE

LEVELNIC DL-S4 series ・DL-S3 series ・DL-S2W series,
equipped with L-type base enables measurement in the Z-axis in addition to X-axis and Y-axis

LEVEL
PROTRACTOR● DIMENSIONSUnits： mm

170（MAGNETIZED RANGE）（30）200
215

MAGNET HOLDER

BASE

255
260

55

Order No.

Model No.

Base Squareness（mm）

Weight

010028DL-SL

Within ± 0.04

4.9kg

127

再 校

02_121-160角度.indd 127

レベルニック

L型ベース

T_341

CMYK

2019/02/26 13:27

19/02/26 13:27:32

USE●L-type base for Levelnic
MATERIAL●Cast iron
FEATURES●Applicable Levelnic：DL-S4W series, DL-S3 series, DL-S2W series●Vertical（Z-axis）measurement is available by applicable Levelnic mounted on●L-type magnetic base enables vertical（Z axis）measurement●With 130°V groove
ACCESSORIES● Shock resistant aluminum case

φ30

MAGNET ON/
OFF LEVER HANDLEANGLELEVELNIC DL-S2W SERIES

PDI

Wireless Data transmission. Same specification as DL-S3, and wireless system equipped.
The maximum longitudinal coverage is 50 ～ 100m.

HOW TO MOVIEFor vertical measurement▶
L-type Base■ High precision Digital Level（Wireless by Bluetooth）■ Resolution 0.001mm/m■ Stainless steel base type that
not requires rust prevention
maintenance（DL-S2W SUS /
DL-S2W（USB）SUS）

P127
For setting a fixed pitch▶
Levelnic Base with Adjustable Pitch

P137
For flatness andstraightness calculation▶
Levelnic Measurement Software

P138
〜139

Rust prevention▶
Zerust Film for SteelUSE●High precision leveling of machine tools and surface plates.
Straightness, flatness and vertical measurement.
MATERIAL●Body：AluminumFEATURES●Wireless data transmission●Wireless type Levelnic, useful for leveling and tilt measurement;
-In overhead location
-In places where the cable can not reach●PC screen becomes display of Levelnic
-Measurement data can be displayed on PC, using receiver and data display software●Custom-made base can be attached to the threaded screw holes（M5 × P0.8）at the bottom base※This model does not have display on the unit. PC or display unit is required for display the measured values.

P380

SPECIFICATIONSLEVELNICPRECISION
LEVEL

Measuring Range± 5.000mm/m・± 0.2865°

Minimum Reading

0.001mm/m・0.0001°

Operating Temperature

0 〜 40℃

Accuracy ※1

Operating temperature 17 〜 23℃	± 0.85％ rdg（0 〜± 2mm/m, 0 〜± 0.1146°）± 1.0％ rdg（± 2 〜± 5mm/m, ± 0.1146 〜± 0.2865°）

Operating temperature 0 〜 40℃	± 2.6％ rdg（0 〜± 2mm/m, 0 〜± 0.1146°
）

ANGLE

LEVEL
PROTRACTOR± 2.7％ rdg（± 2 〜± 5mm/m, ± 0.1146 〜± 0.2865°）

Repeatability

Within ± 0.005mm/m・± 0.0003°

Radio System

Bluetooth Class1 ※2

Power Source

9V 006P type dry cell 1 pc

（nominal voltage 8.4V rechargeable battery can also be used）
AC adapter（DC9V output ・AC100V ～ 240V）included

Run time（Continuous use）

Alkaline dry cell： Approx. 10 hrs（nominal voltage 8.4V rechargeable battery： Approx. 3 〜 4 hrs.）※ May vary depending upon measurement conditions

Dimensions

L173 × W59 × H147mm（H： up to top of antenna）（base dimensions： L150 × W55mm）
DL-S2W, DL-S2W SUSDL-S2W-E● Shock resistant aluminum case● Warranty・Instruction manual・Inspection report●AC adapter（DC9V output・AC100V ～ 240V）● Power source cable for wireless unit

Accessories
DL-S2W（USB）SUS
DL-S2W（USB）● Shock resistant aluminum case

● Receiving unit ※3● Warranty・Instruction manual・Inspection report●AC adapter（DC9V output・AC100V ～ 240V）● USB receiving unit

●9V dry cell

●Dedicated display software●9V dry cell

●Dedicated display software※ 1： % rdg indicates percentage of displayed value※ 2： Our Bluetooth system qualification is with TELEC（Japan）, FCC（U.S.A）, CE（EU）, KCC（South Korea）※ 3： PC without D-USB pin port（RS-232C）requires USB serial transmitter（optional）

19/02/26 14:47:46

Order No.

010017

Model No.

DL-S2W

Receiving Unit Type

Software Language

D-SUB9010054DL-S2W（USB）

USB

010114DL-S2W SUSD-SUB9010115

DL-S2W（USB）SUS

010029DL-S2W-E

Cast ironJapanese
Stainless steel SUS303

USB
D-SUB9Base Material

EnglishCast iron

Weight

2.1kg
2.1kg
2.1kg
2.1kg
2.1kg

128

再 校

02_121-160角度.indd 128

CMYK

2019/02/26 14:47

PDI

Digital display with 0.01mm/m resolution.  Unlike bubble level, there is no delay
waiting for reading to settle, reducing human error.

■ High precision Digital Level■ Resolution 0.01mm/m

SPECIFICATIONS± 5.00mm/m

Minimum Reading

0.01mm/m

For vertical measurement▶
L-type Base■ Stainless steel base type（DL-m1 SUS）

USE
●High precision leveling of machine tools and surface plates.
Straightness and flatness measurement.
MATERIAL●Body：AluminumFEATURES●Compact, lightweight and simple operation●Easy to fix the datum point, divide measurement
value into half with one touch operation●Fast output reduces installation time for field work
Measuring Range

HOW TO MOVIEANGLELEVELNIC DL-m1 SERIES

P130
Rust prevention▶
Zerust Film for SteelP380

DL-m1 SUS
Order No.

Model No.

Base Material

Weight

010014

DL-m1

Cast iron

880g

010117DL-m1 SUS

Stainless steel SUS303

920g

Operating Temperature

0 〜 40℃

Accuracy ※Operating temperature 17 〜 23℃ ± 0.02mm/m or ± 3% rdg whichever is larger（0 〜± 2mm/m）± 4% rdg（± 2 〜± 5mm/m）
Operating temperature 0 〜 40℃ ± 0.02mm/m or ± 5% rdg whichever is larger（0 〜± 2mm/m）± 6% rdg（± 2 〜± 5mm/m）

Repeatability

Within ± 0.01mm/m

Power Source

9V dry cell 1 pc

Run time（continuous use）

Zinc-carbon cell： Approx. 45 hrs. / Alkaline cell： Approx. 90 hrs. ※ May vary depending upon the measurement conditions

Dimensions

L105 × W50 × H73mm（base dimensions： L100 × W50mm）

Accessories● Shock resistant aluminum case● Warranty・Instruction manual・Inspection report● 9V dry cell

※ % rdg indicates the percentage of displayed value
LEVELNICLEVELNIC DL-m3 SERIESPDIDigital display with 0.01mm/m resolution.  Unlike bubble level, there is no delay
waiting for reading to settle, reducing human error.

■ High precision Digital Level■ Resolution 0.01mm/m

HOW TO MOVIEFor vertical measurement▶
L-type Base■ Stainless steel base type（DL-m3 SUS）

P130

PRECISION
LEVEL
ANGLE

LEVEL

For flatness andstraightness calculation▶

With data output

Levelnic Measurement Software

USE●High precision leveling of machine tools and surface plates.
Straightness and flatness measurement.
MATERIAL●Body：AluminumFEATURES●Digital display for easy to read measurement●Compact, lightweight and simple operation●Equipped with RS-232C output for combination use of Data
recording, Straighness and Flatness software
SPECIFICATIONS
Measuring Range± 5.00mm/m

Minimum Reading

0.01mm/m

P138
〜139

PROTRA
CTOR

Rust prevention▶
Zerust Film for SteelP380

DL-m3 SUS
Order No.

Model No.

Base Material

Weight

010015

DL-m3

Cast iron

880g

010118DL-m3 SUS

Stainless steel SUS303

920g

Operating Temperature

0 〜 40℃

Accuracy ※1

Operating temperature 17〜23℃	±0.02mm/m or ±3% rdg whichever is larger（0〜±2mm/m） ±4% rdg（±2〜±5mm/m）
Operating temperature 0〜40℃ ±0.02mm/m or ±5% rdg whichever is larger（0〜±2mm/m） ±6% rdg（±2〜±5mm/m）

Repeatability

Within ± 0.01mm/m

Output SIgnal

Conforms to RS-232C ※2
9V dry cell 1 pc

Run time（continuous use）

Zinc-carbon cell： Approx. 35 hrs. / Alkaline cell： Approx. 70 hrs. ※ May vary depending upon measurement conditions

Dimensions

L105 × W50 × H73mm（base dimensions： L100 × W50mm）

Accessories● Shock resistant aluminum case● Warranty・Instruction manual・Inspection report● 9V dry cell

※ 1： % rdg indicates percentage of displayed value※ 2： PC without D-USB pin port（RS-232C）requires USB serial transmitter（optional）

129

再 校

02_121-160角度.indd 129

CMYK

2019/02/26 13:09

19/02/26 13:09:07

Power Source

ANGLELEVELNIC DL-m3LS

PDI

Levelnic DL-m3 on L-type Base.
X-axis, Y-axis as well as Z-axis measurements are possible.

HOW TO MOVIEFor flatness andstraightness calculation▶
Levelnic Measurement Software

P138
〜139

Rust prevention▶
Zerust Film for Steel

P380
USE
●High precision leveling of machine tools and surface plates.
Straightness, flatness and vertical measurement.
MATERIAL●DL-m3 / Body：Aluminum●L-type base：Cast iron
FEATURES●Digital display●Compact and lightweight●Equipped with RS-232C output for combination use of Data
※ Base does not have magnetic capability
recording, Straighness and Flatness software●L-type base allows measurements in the vertical direction●With 130°V groove
ACCESSORIES●Shock resistant aluminum case
●Warranty・Instruction manual・Inspection report●9V dry cell
Please refer to DL-m3（P129）for detailed specifications for this unit
Order No.

Model No.

Base Squareness（mm）

Weight

010019DL-m3LS

Within ± 0.035

2.8kg

LEVELNICPRECISION
LEVEL
ANGLE

L-TYPE BASE FOR LEVELNIC

DL-mL

Rust prevention▶
Zerust Film for SteelLevelnic DL-m1, m3 mounted on base will be able to take vertical measurements

P380

LEVEL
PROTRACTOR● DIMENSIONSUnits： mm+$1'/(

19/02/26 14:47:50+$1'/(

USE●L-type base for Levelnic
MATERIAL●Cast iron
FEATURES●Applicable Levelnic：DL-m1 series, DL-m3 series●Vertical（Z-axis）measurement is available by applicable Levelnic mounted on●With 130°V groove
ACCESSORIES●Shock resistant aluminum case

/6+$3(%$6(%8775(663/$7(

5(7(17,21.12%※ Base does not have magnetic capability

Order No.

Model No.

Base Squareness（mm）

Weight

010022DL-mL

Within ± 0.035

1.9kg

130

再 校

02_121-160角度.indd 130

CMYK

2019/02/26 14:47

For flatness andstraightness calculation▶
Levelnic Measurement Software

Attach Bluetooth Box to applicable levelnic for wireless data transmitting.
Can be functioned in any countries by using Bluetooth dongle.

P138
〜139

USE
●Wireless data transmitting is available by 
attaching to applicable Levelnics
MATERIAL●Body：Aluminum●Switch Panel：ABS
FEATURES●RS-232C output signal from Levelnic is transmitted by Bluetooth●Can be used in any country with a locally available dongle●Eliminating cables greatly improves work efficiency

ANGLE

BLUETOOTH BOX

Mounting example to DL-S3

SPECIFICATIONS
Input

Levelnic RS-232C signal only

Wireless

Bluetooth ※ Class depends on USB dongle

Applicable USB
Dongle adaptor

Bluetooth 3.0 and higher, SPP profile

Power Source

Lithium polymer battery（chargeable with USB charging cable）

Run time（continuous use）

Approx. 7 hrs.

Dimensions

L82 × W35 × H24mm

Charging

Approx. 2 hrs.

Applicable Levelnic

DL-S3 series, DL-m3 series, DL-m3LS, DL-D3, DL-S3L

Accessories

Data cable, USB charging cable, mounting plate, instruction manual, Bluetooth USB dongle adaptor Class 1（Japan Only）

Order No.

Model No.

Weight152036

DTW-DL0180g
LEVELNICLEVELNIC WIRELESS ADAPTER

DL-BW / DL-BW（USB）

PRECISION
LEVEL

Allows communicating with Levelnic using Bluetooth.  
Wireless range of 50-100mm line-of-sight.

ANGLE

LEVEL
PROTRACTOR

USE●Wireless adapter for Levelnic
MATERIAL●Body：Stainless
FEATURES●The RS232C output signal is transmitted using Bluetooth
wireless●Eliminating cables greatly improves work efficiency

Mounting example to DL-S3C

SPECIFICATIONS
Input

Levelnic RS-232C signals（Levlnic use only）

Wireless

Bluetooth Class1 ※1

Power Source

AA battery 6 pcs（rechargeable batteries can be used）

Run time（continuous use）

Alkaline cell： Approx. 70 hrs. / Zinc-carbon cell： Approx. 30 hrs. 1700mAh rechargeable battery： Approx. 45 hrs.
※ May vary depending upon measurement conditions

Dimensions

L143 × W50 × H173mm（including trasmission unit）

Applicable Levelnic

DL-S3 series, DL-S3L, DL-m3, DL-m3LS, DL-D3※ DL-S3 series can direct mount

Accessories

Mounting parts, receiving unit, USB power cable for receiving unit（DL-BW only）, AA battery × 6, instruction manual

Order No.

Model No.

010038DL-BW

010055

DL-BW（USB）

Receiving Unit TypeD-SUB9USB※2

Weight

530g
530g

131

再 校

02_121-160角度.indd 131

CMYK

2019/02/26 13:09

19/02/26 13:09:14※ A small thermal printer can not be used
※ 1： Our Bluetooth system qualification is with TELEC（Japan）, FCC（U.S.A）, CE（EU）, KCC（South Korea）※ 2： PC without D-USB pin port（RS-232C）requires USB serial transmitter（optional）

ANGLE

TRANSMITTER / DISPLAY SET
For displaying measurements from two DL-S3 series products in one handy device at the same time.
Suitable for X and Y direction measurement.
USE●Transmit and display measurements 
for up to two Levelnic DL-S3 series products
FEATURES●Simultaneous display of two connected DL-S3 series products. 
Useful for measuring X and Y directions.
●Eases installation of large machine tools by 
allowing multiple operators to work at same time●Measured data is more easily visible, especially when used in locations which are remote or have low clearance height●Since transmitter and receiver are a paired set, there is no needto set up after purchase -- it can be used immediately

Connect DL-S3 toDL-D5 with cable

Transmitter sends data to display unit
using Bluetooth Class 1 wirelessOne display unit can display measured
value for two connections

50̃100m line of sight
LEVELNIC
DL-S3

DL-D5 (Display)

DL-D5(Transmitter)

・With Back-lite
・mm/m display

It is possible to connect up to
two DL-S3 to a single transmitter.

Independent 0-cal. and 1/2-cal.
capability for each display

Can also
Useful for measuring
be used with
X and Y directionat same time. single
DL-S3

SPECIFICATIONSTransmitter
Input

Display

Levelnic DL-S3 series external signal（×2）

Display−
16 char × 2 linescharacter size = 4.9 × 2.5mm with backlight−

Wireless

Bluetooth Class1

Communication Range

50 ～ 100m line-of-sight（depending on conditions）

Operating Temperature

0 ～ 40℃

Power

2 × AA battery（can also use commercially available rechargeable batteries）※1

Alkaline cells： Approx. 8 hrs.
Communicating with display device： Approx. 17 hrs.
attempting communication with display： Approx. 3 hrs.
144 × W76 × D27mm（including lanyard ring）
Dimensions（when mounted in silicone cover： L147 × W81 × D35mm）
2 × connecting cable（approx. 80cm length）
Accessories
2 × AA battery, silicone cover
2 × AA battery, silicone cover※ 1： Our Bluetooth system qualification is with TELEC（Japan）, FCC（U.S.A）, CE（EU）, KCC（South Korea）※ 2： The value of three decimal places digit of the display may show inappropriate number when using DL-m3※ 3： Including batteries and silicone coverOperating life
（continuous use）

LEVELNICPRECISION
LEVEL

Order No.

Model No.

010043DL-D5

Compatible Model

DL-S3 series・DL-S3L・DL-m3

Weight290g each ※ 3

※2

ANGLE

LEVEL
PROTRACTOR

LEVELNIC DISPLAY
Can display data at a distance. Enables measurement process to be done by one person.
The work for two can be done by one person.
USE●Levelnic 
measurement data display
MATERIAL●Body：Stainless steel
FEATURES●Can divide measurement value into half with 
one-touch operation（DL-S2W series only）

19/02/26 13:09:18

SPECIFICATIONS

Combination example with DL-S2W
Order No.

Model No.

Weight

010024

DL-D4

330g

Display Range

As per Levelnic spec.

Minimum Reading

As per Levelnic spec.

Operating Temperature

0 〜 40℃

Input Signal

Conforms to RS-232C（Levelnic digital output signal only）

Power Source

9V dry cell（nominal voltage 8.4V rechargeable battery can also be used）/ AC adapter（included）

Run time（continuous use）

With wireless connection： Alkaline cell： Approx. 5 hrs.
With cable connection： Alkaline cell： Approx. 60 hrs.
※ May vary depending upon measurement conditions

Dimensions

L116 × W62 × H48mm

Applicable Levelnic

DL-S3 series, DL-S3L, DL-m3 series, DL-m3LS, DL-S2W series, DL-S4W series（wired）※ DL-P5AD, DL-P6 can also use cable connection※ DL-S2W, DL-BW can connect using buit-in wireless capability※ DL-S4W（USB receiving unit）, DL-S2W（USB）, DL-BW（USB）cannot be used on this unit

Accessories

AC adaptor（DC9V output・AC100V ～ 240V）, 9V dry cell, instruction manual

132

再 校

02_121-160角度.indd 132

CMYK

2019/02/26 13:09

ANGLELEVELNIC DATA SYSTEM
Enables data display and saving of up to eight Levelnic with one PC

USE●For data management of multiple Levelnics
MATERIAL●Body：Stainless steel
FEATURES●Eight Levelnics can be connected at once
●Data display & recording software included（auto capture function）

Example using multiple units
with Bluetooth wireless communication

SPECIFICATIONS

Order No.

Model No.

Software Language

Weight

010037

DL-ADP

Japanese

1.6kg

010089DL-ADP-E

English

1.6kg

Input

D-SUB（9 pins, male）
RS-232C compliant, 8 inputs（for Levelnic connection only）
1200 baud, 1 stop bit, no parity, no handshaking）

Output※
D-SUB（9 pins, male）
RS-232C compliant, 1 output（PC connection）
9600 baud, 1 stop bit, no parity
Hardware flow control

Operating Temperature

0 〜 40℃

Power Source

AC100V adapter（DC9V output）
AA batteries 6 pcs（rechargeable battery can also be used）※ Run time depends on number of Levelnics

Dimensions

W180 × H150 × D130mm

Applicable Levelnic

DL-S3 series, DL-S3L, DL-m3 series, DL-m3LS, DL-S2W series, DL-S4W series（wired）※ DL-P5AD, DL-P6 can also use cable connection※ DL-S2W, DL-BW can connect using buit-in wireless capability※ DL-S4W（USB receiving unit）, DL-S2W（USB）, DL-BW（USB）cannot be used on this unit

Accessories

Data recording software, RS-232C cross cable, AC adaptor（DC9V output 100V）, AA battery × 6, instruction manual

LEVELNIC※ PC without D-USB pin port（RS-232C）requires USB serial transmitter（optional）

PRECISION
LEVEL
ANGLE

LEVEL
PROTRACTOR

LEVELNIC SD LOGGER 3
Measured value is recorded directly on SD card.
Loading data automatically at a specific interval by script function.
USE●To record measured value by Levelnic●Measured value can be recorded on SD card without PC
FEATURES●To record data on sites that PC is not suitable●Recorded data by text format can be used with other software●Loading data automatically at a specific interval by 
script written in advance on SD card
●Paperless by saving data on SD card
SPECIFICATIONS
Levelnic RS-232C signal

SD card

SD / SDHC Max. 32GBPower

AC adaper（DC9V）included

Dimensions

W120 × D100 × H39mm

Applicable Levelnic

DL-S4W（wired）, DL-S3 series, DL-S3L, DLm3 series, DL-m3LS, DL-D3

Order No.

Model No.

Weight

010044DT-MCK3-XS-SK

170g

133

再 校

02_121-160角度.indd 133

CMYK

2019/02/26 13:09

19/02/26 13:09:20

Input

ANGLESEPARATE SYSTEM / LEVELNIC CABLE

PDI■ High accuracy Digital Level■ Resolution 0.001mm/m

HOW TO MOVIE

DL-P1
●

DL-P3

Digital sensor（DL-S2）・Digital display（DL-D3）・
Cable（DL-P1）as combined system※ PC is not included([DPSOHRIFRQQHFWLQJPXOWLSOHV\VWHPV6HSDUDWHV\VWHP

&RQQHFWLRQFDEOH '/3
3&

6HSDUDWHV\VWHP
DL-S2

DL-D3

&RQQHFWLRQFDEOH '/3

6HSDUDWHV\VWHP

USE●High precision leveling of machine tools and surface plates.  Straightness and flatness measurement for digital display.
MATERIAL●Body：Aluminum●Sensor base：Cast iron
FEATURES●Display allows machine tool data obtained by Levelnic to be close at hand
●Output data is available on built in RS-232C interface for connection to computer or printer●Multiple output signals can be combined onto one signal allowing expansion to 2, 3,  or 4 devices for easy processing●Easy to fix the Datum point, divide measurement value into half with one touch operation●Custom-made base can be attached to the threaded screw holes（M5 × P0.8）at the bottom base
SPECIFICATIONSLEVELNICPRECISION
LEVEL
ANGLE

LEVEL
PROTRACTOR

Measuring Range± 5.000mm/m・± 0.2865°

Minimum Reading

0.001mm/m・0.0001°
0.01mm/m・0.001°

Operating Temperature

0 〜 40℃

Accuracy

Operating temperature 17 〜 23℃ ± 0.85％ rdg（0 〜± 2mm/m・0 〜± 0.1146°）± 1.0％ rdg（± 2 〜± 5mm/m・± 0.1146 〜± 0.2865°）
Operating temperature 0 〜 40℃
± 2.6％ rdg（0 〜± 2mm/m・0 〜± 0.1146°）± 2.7％ rdg（± 2 〜± 5mm/m・± 0.1146 〜± 0.2865°）

Repeatability

Within ± 0.005mm/m・± 0.0003°

Output SIgnal

Conforms to RS-232C ※Power Source

AA battery 6 pcs
AC adapter（DC9V output・AC100V ～ 240V）included

Run time（continuous use）

Zinc-carbon cell： Approx. 27 hrs.
Alkaline cell： Approx. 55 hrs.
※ May vary depending upon measurement conditions

Dimensions

DL-S2 L167 × W59 × H111mm（base dimension： L150 × W55mm）
DL-D3 L235 × W106 × H51mm
DL-P1 Length： 5m（standard cable）／ DL-P3 Length： 4.5m

Accessories● Shock resistant aluminum case● Warranty・Instruction manual・Inspection report●AC adapter（DC9V output・AC100V ～ 240V）※ PC without D-USB pin port（RS-232C）requires USB serial transmitter（optional）
Order No.

Product

Model No.

010023DISPLAY UNIT

DL-D3

Weight1kg

010012LEVELNIC

DL-S2

2kg

010031CONNECTING CABLE FOR LEVELNIC

DL-P1

400g

010033

CONNECTING CABLE FOR DISPLAY UNIT

DL-P3

100g

19/02/26 13:09:23■ PC connection cable

USE●For connecting Levelnic to PC
SPECIFICATIONS●Length：φ3mm×5m●9 pins connector（serial port connection）●Applicable Levelnic：DL-S4W series, DL-S3 series, DL-S3L, DL-m3 series, DL-m3L, Separate system※ PC without D-USB pin port（RS-232C）requires USB serial transmitter（optional）
Order No.

Model No.

Material

010040DL-P6

2 cond. w/ shield（Cu）

134

再 校

02_121-160角度.indd 134

CMYK

2019/02/26 13:09

DL-S1PDI

Digital display with 0.01mm/m resolution.
Unlike bubble level, there is no delay waiting for reading to settle, reducing human error.

HOW TO MOVIEANGLELEVELNIC■ High precision Digital Level■ Resolution 0.01mm/m

USE●High precision leveling of machine tools and surface plates, straightness and flatness measurement
MATERIAL●Body：Aluminum●Base：Cast iron
FEATURES●Faster response than bubble level●Wider range of measurement than conventional level●Measurement can be taken in difficult-to-see position by pressing hold switch and then reading

LEVELNICPRECISION
LEVEL
ANGLE

SPECIFICATIONS
Measuring Range± 9.99mm/m

Minimum Reading

0.01mm/m

Operating Temperature

0 〜 40℃

Accuracy ※1

Operating temperature 17 〜 23℃± 1.3% rdg

Operating temperature 0 〜 40℃± 3.0% rdg

Repeatability

Within ± 0.01mm/m

Power Source ※2

9V dry cell 1 pc

LEVEL
PROTRACTOR

AC adapter（100V 〜 240V）（optional）
Run time（continuous use）

Zinc-carbon cell： Approx. 25 hrs.
Alkaline cell： Approx. 50 hrs.
※ May vary depending upon measurement conditions

Dimensions

L159 × W78 × H104mm（base dimension： L150 × W55mm）

Accessories● Shock resistant aluminum case
● 9V dry cell

● Warranty・Instruction manual・Inspection report※ 1： % rdg indicates percentage of displayed value※ 2： AC adapter sold separately
Order No.

Model No.

Weight

010011

DL-S1

2.0kg

Model

DL-S1V

Rolling Error

Within 0.1mm/m when the bubble of auxiliary bubble tube is centered

Plane and V groove error

Within 0.05mm/m

V groove angle

130°（for cylindrical objects up to φ 65mm ）

135

再 校

02_121-160角度.indd 135

CMYK

2019/02/26 13:09

19/02/26 13:09:26

With V groove base - available on requestANGLELEVELNIC EL-01A SERIESPDIHigh-sensitive（0.01mm/m）. Pendulum type level with analog output.
Three levels of sensitivity.

■ High precision Analog Level■ Resolution 0.01mm/m

USE●Leveling of machine tools and surface plates. 
Straightness and flatness measurements.
MATERIAL●Body：AluminumFEATURES●Quicker response than bubble level●Sensitivity is switchable in three steps●Connectible directly to commonly-available analog recorder throughanalog output●Custom-made base can be attached to the threaded screw holes （M5 × P0.8）at the bottom base●Pendulum system prevents any hysteresis errors and is not
influenced by temperature change
SPECIFICATIONS
Sensitivity

Ⅰ： 1Division ＝ 0.01mm/m
Ⅱ： 1Division ＝ 0.05mm/m
Ⅲ： 1Division ＝ 0.1mm/m

LEVELNICPRECISION
LEVEL
ANGLE

LEVEL

Measuring Range± 0.2mm/m（1Division 0.01mm/m）（3 Range）± 1mm/m（1Division 0.05mm/m）± 2mm/m（1Division 0.1mm/m）

Operating Temperature

0 〜 40℃

Pointing Error

Within ± 0.5 Division

Repeatability Accuracy

Within ± 0.005mm/m

Output SIgnal

Analog output± 200mV F.S.（for all range）

PROTRACTOR

Connectible load resistance： No less than 100 ΩPower Source

9V dry cell 1 pc
AC adapter（DC9V output ・AC100V ～ 240V）included

Run time（continuous use）

Zinc-carbon cell： Approx. 25 hrs.
Alkaline cell： Approx. 50 hrs.
※ May vary depending upon measurement condition

Base Dimension

L150 × W85 × H195mm※ Excluding Handle（dimension： L150 × W55mm）

Accessories● Shock resistant aluminum case● Warranty・Instruction manual・Inspection report● AC adapter（DC9V output・AC100V ～ 240V）
Order No.

Model No.

010005EL-01A

010116EL-01A SUS● 9V dry cell

Base Material

WeightCast iron

2.9kg

Stainless steeel SUS303

2.9kg

19/02/26 13:09:28■ Connection Cable / EL-K2
USE●Connects to analog recorder with analog output of EL-01A
SPECIFICATIONS●Length：φ3 × 5m ●Recorder：Banana plug●Levelnic：M3 plug
Order No.

Model No.

Weight

010082

EL-K2

150g

136

再 校

02_121-160角度.indd 136

CMYK

2019/02/26 13:09

ANGLELEVELNIC BASE WITH ADJUSTABLE PITCH
By attaching Levelnic to this base, any desired measurement pitch can be set● DIMENSIONS+$1'/(

LP-300USE*$8*(%/2&.
●Setting a fixed pitch for straightness and flatness measurements
MATERIAL●Carbon steel for machine structural use
FEATURES●Applicable levelnic：DL-S4 series, DL-S3 series, DL-S2W series, DL-S2, EL-01A series●The base is graduated in mm and enables setting measurement pitch●Measuring face consists of 2 pcs of 15mm Gauge block*$8*(%/2&.
Units： mm
Model No.

A

B

C

D

LP-200

230

180

65

55

LP-300

330

180

65

55

Order No.

Model No.

Measurement pitch range（mm）

Weight

010020LP-20050 〜 200

1.8kg

010021LP-30050 〜 300

2kg
LEVELNICPRECISIONLEVELCARRYING CASE FOR LEVELNIC
Shock resistant aluminum case for Levelnic.  Suitable for storage and carrying.

ANGLE

LEVEL
PROTRACTOR

LV-CM1Model No.

Dimensions（mm）

010025

LV-CS3362 × 236 × 184

1.50kg010026

LV-CM1250 × 200 × 150

0.97kgOrder No.

Weight137

再 校

02_121-160角度.indd 137

CMYK

2019/02/26 13:09

19/02/26 13:09:30

USE
LV-CS3●Carrying case for Levelnic
（Levelnic is a precision instrument, we recommend that you use a carrying case）
MATERIAL●Body：Aluminum●Inside：Urethane foam
FEATURES●Applicable levelnic：
LV-CS3：DL-S4W series, DL-S3 series, DL-S2W serires, DL-S2, DL-S1
LV-CM1：DL-m1 series, DL-m3 series●Shock protection during transport

ANGLESOFTWARE FOR LEVELNIC
Software for Levelnic.
Simple operation and graphics for easy use and understanding.
Straightness measurement computer software（ST）
Flatness measurement computer software（FL）■ Greatly reduces measurement time by recording the results for straightness and flatness measurements■ For cabled and wireless system■ Supported OS： Windows XP, Vista, 7, 8, 10■Direct input from DL-S4W series, DL-S3 series, DL-S3L, DL-S2W series, DL-S2+D3, DL-m3 series, DL-m3LS, and Separate System■ Flatness measurement■ Straightness measurement while connected to PC
Connection cableOR
Separate System

DL-P5AD、
DL-P6

PC

Printer

Software forlevelnicOS、DL-S4W

wirelessversion does
not require
cable●Items within doted line to be supplied by user

No need to install the software to PC

Only drag and drop EXE file, and double click to start. Also can be used directly from USB memory.

Input selection

Data can be manually typed in from any level, including bubble tube type level

Input signal from Levelnic over RS-232C

With computer connected to the Levelnic output, sequential measurements taken at regular intervals can be used to calculate flatness
and straightness for display, printing and storage.（saved measurement data can not be revised）
LEVELNICPRECISION
LEVEL
ANGLE

Input typed onto keyboard from bubble level for display, calculation and storage

Using a bubble level and keyboard to enter data, sequential measurement taken at regular intervals can be used for calculation, display
and storage

Displacement measurements（µm）typed onto keyboard for display, calculation and storage

Measurement of displacement from a reference plane made with a linear gauge or dial indicator at regular intervals can be used in placeof an electronic angle gauge（goniometer）and entered into keyboard for calculation, display and storage

LEVEL
PROTRACTOR

LEVELNIC MEASUREMENT SOFTWARE
Use to easily calculate straightness when connect to Levelnic with cable or wireless connection

HOW TO MOVIE■ Straightness software● With Straightness software Levelnic can easily perform the straightness measurements that once required an autocollimator, and for a lower cost

19/02/26 13:09:36■ Straightness measurement

USE●Software to perform straightness calculations
FEATURES●Both wired and wireless capability●Model compatibility：DL-S4W series, DL-S3 series, DL-S3L, DL-S2W series, DL-S2＋D3, DL-m3 series, DL-m3LS
※Above compatibility applies for direct connection. Data can be manually typed in from any model, including bubble tube type level.
SPECIFICATIONS●Supports：Windows XP, Vista, 7, 8, 10
Order No.

Model No.

Language

010066

ST

Japanese

010070ST-E

English010073ST-C

Chinese138

再 校

02_121-160角度.indd 138

CMYK

2019/02/26 13:09

Use to easily calculate flatness when connect to Levelnic with cable or wireless connection

HOW TO MOVIEANGLELEVELNIC MEASUREMENT SOFTWARE■ Flatness software● With Flatness software Levelnic can easily perform the flatness measurements that once required an autocollimator, and for a lower cost
■ Flatness measurement

USE●Software to perform flatness calculations
FEATURES●Both wired and wireless capability●Model compatibility：DL-S4W series, DL-S3 series, DL-S3L, DL-S2W series, DL-S2＋D3, DL-m3 series, DL-m3LS
※Above compatibility applies for direct connection. Data can be manually typed in from any model, including bubble tube type level.
SPECIFICATIONS●Supports：Windows XP, Vista, 7, 8, 10
Order No.

Model No.

Language

010065

FL

Japanese

010069

FL-E

English010072FL-C

ChineseLEVELNIC DATA RECORDING SOFTWARELEVELNIC■ Data recording software LEVELOG
USE●Records output data from Levelnic
FEATURES●Levelnic measurements are recorded to text file
●Model compatibility：DL-S4W series, DL-S3 series, DL-S3L, 
DL-S2W series, DL-S2＋D3, DL-m3 series, DL-m3LS
SPECIFICATIONS●Supports：Windows XP, Vista, 7, 8, 10

PRECISION
LEVEL
ANGLE
Order No.

Model No.

Language

010067LEVELOG

Japanese

010068LEVELOG-E

EnglishLEVELPROTRACTOR■ Data recording software LEVELOG-M

For data management of multiple use of Levelnic.
Display and record up to eight data channels.

Order No.

Model No.

Language

010074LEVELOG-M

Japanese

010075LEVELOG-M-E

English139

再 校

02_121-160角度.indd 139

CMYK

2019/02/26 13:09

19/02/26 13:09:38

USE●For displaying measurement values on PC screen and loading its data into PC for up to eight units of Levelnic
FEATURES●Loadable up to 14,400 data points which can be saved as plain text
●Selectable data loading - Manual loading / Auto-loading at set time interval●Model compatibility：DL-S4W series, DL-S3 series, DL-S2W series, DL-S2+DL-D3, DL-m3 series
SPECIFICATIONS●Supports：Windows XP, Vista, 7, 8, 10※ Computer requires COM port or USB port for each connected Levelnic