KSPECT/MULTI SPECT取り扱い説明書
Rev1.0改定 No
初版

改定内容頁

目次
KSPECT/MULTI SPECT ........................................................................................................................... 1
1.はじめに .................................................................................................................................................... 11）対象製品： .......................................................................................................................................... 12）動作環境： .......................................................................................................................................... 13）ソフトウエアの入手方法 .................................................................................................................... 14）ソフトウエアのインストール ............................................................................................................. 15）ソフトウエアの起動 ............................................................................................................................ 12.画面............................................................................................................................................................ 2
3.制御画面..................................................................................................................................................... 31）Acquisition タブ................................................................................................................................... 3a)スタート/ストップ ............................................................................................................................ 3b)データ消去......................................................................................................................................... 3c)プリセット測定 .................................................................................................................................. 3d)エネルギー較正 ................................................................................................................................. 3e)繰り返し測定の設定 .......................................................................................................................... 42) Spectra タブ .......................................................................................................................................... 5a)カウント軸スケールの選択 ............................................................................................................... 5b)チャネル軸表示範囲の選択 ............................................................................................................... 6c)ROI 設定 ............................................................................................................................................ 7d) エネルギー表示モード（MSA/MSP のみ） ................................................................................... 83) Analysis タブ （MSA/MSP のみ） .................................................................................................... 94.)Setup タブ........................................................................................................................................... 11a)

General タブ ................................................................................................................................ 11b)

エネルギー較正タブ .................................................................................................................... 11c)

エネルギー較正タブ(MSA/MSP のみ) ....................................................................................... 12d)

効率較正タブ(MSP のみ) ............................................................................................................ 12e)

測定情報の表示............................................................................................................................ 14f)

制御対象の選択（KSpect のみ） ................................................................................................. 155) Report タブ ......................................................................................................................................... 15
6.File タブ ................................................................................................................................................... 16
B.メイン画面 .............................................................................................................................................. 17
1.

Spectra タブ......................................................................................................................................... 17

2.

Monitor タブ (MSA/MSP のみ) ......................................................................................................... 18

C.情報画面 .................................................................................................................................................. 19
D ステータス画面 ....................................................................................................................................... 20

Appendix 1 スペクトルデータフォーマット .............................................................................................. 21
Appendix 2 .................................................................................................................................................. 26

図表目次

Figure 1 メイン画面 ........................................................................................................................... 2
Figure 2 エネルギー較正画面 ............................................................................................................. 4
Figure 3 繰り返し測定の設定 ............................................................................................................. 4
Figure 4Spctra タブ ........................................................................................................................... 5

Figure 5 カウント軸操作 .................................................................................................................... 5
Figure 6 チャネル軸操作 .................................................................................................................... 6
Figure 7 拡大表示時の画面 ................................................................................................................ 6
Figure 8

ROI 設定 .............................................................................................................................. 7

Figure 9

ROI 設定時の画面表示......................................................................................................... 7

Figure 10 ROI 情報表示 ..................................................................................................................... 8
Figure 11 エネルギー表示モード ....................................................................................................... 8

KSPECT/MULTI SPECT 取り扱い説明書

1.はじめに
本書は英国 KromeK 社が製造/仁木工芸株式会社が販売する以下の製品の取り扱いについて記載されています。より詳細はメーカ発行のマニュアル（英文）を参照してください。英文マニュアルはプログラムの HELP から入手できます。
1）対象製品：
KSpect （以下 KSP と記載）

無償ダウンロードソフト

Multi-Spect Analysis ( 以下MSA と記載)

有償ダウンロード・アクティベーション

Multi-Spect Premium (以下MSP と記載)

有償ダウンロード・アクティベーション

102）動作環境：
対象製品は Microsoft 社 Windows7/Windows8/Windows101がインストールされ、オペレーティングシステムが正常に動作する PC にインストールしてご使用ください。
インストール実施時には必ずインターネットに接続できる環境でご使用ください。KSP/MSA/MSP はイ
ンストール時にインターネットから必要なファイルをダウンロードしてインストールを継続します。
3）ソフトウエアの入手方法
KSP/MSA/MSP の各ソフトウエアは KromeK 社ホームページからダウンロードして入手してください。
KromeK 社ホームページは http://www.kromek.com です。
204）ソフトウエアのインストール
ダウンロードしたインストーラは zip 形式で圧縮されているので、PC のハードディスク上に解凍してください。
解凍されたディレクトリにある Setup を実行してインストールを開始してください。
インストール中、KromeK 社製品との通信に必要なドライバ、測定データのデータベース管理に必要なデータベースエンジンをインターネットからダウンロードしてインストールします。
詳しくはインストール中に表示される画面の指示に従ってください。
インストールが正常に終了するとデスクトップに以下のアイコンが表示されます。
305）ソフトウエアの起動
インストールでデスクトップに作られたアイコンをダブルクリックしてプログラムを起動します。

1

Windows7/Windows8/Windows10 は米国 Microsoft 社の登録商標です。

1

KSPECT/MULTI SPECT 取り扱い説明書

2.画面

10

Figure 1 メイン画面

20

A ： 制御画面
ここには PC に接続された KromeK 社製品の測定制御、エネルギー較正、プリセット、リピート測定のコントロールと設定が配置されています。
B ： メイン画面
ここにはスペクトルグラフが表示されます。
C ： 情報画面
30

ここにはスペクトル情報、カーソール位置情報、ROI 情報、設定情報が表示されます。
D ： ステータス画面
ここには PC に接続された KromeK 社のデバイスと状態が表示されます。

2

KSPECT/MULTI SPECT 取り扱い説明書

3.制御画面1）Acquisition タブa)スタート/ストップ
測定を開始するためには10をクリックします。

測定開始中はボタンがに変わります。任意のタイミングで測定を停止するときに使用します。

b)データ消去
データを消去する時にはをクリックします。

c)プリセット測定
20

指定時間の測定を実施する場合には、プリセット値（Real Time/Live Time）をセットします。
単位は秒です。０の場合にはプリセットは無効です。
Real Time でプリセットする場合にはReal Time の欄に数字を入力し、Live Time を０にします。

d)エネルギー較正
エネルギー較正を実施する場合はをクリックします。

30
Figure２ エネルギー較正画面が開くので Channel 番号と Peak Energy のペアを最低２つ入力してください。
２点以上の入力が完了するとフィッティングのグラフが表示されます。
多点入力した際に、入力点がフィッティング線から著しく離れている場合には入力した値を見直してください。

3

KSPECT/MULTI SPECT 取り扱い説明書

Figure 2 エネルギー較正画面e)繰り返し測定の設定
KSP/MSA/MSP では、プリセットで指定された時間で測定を繰り返し実行することができます。
Figure3 繰り返し測定の設定画面にて 繰り返しか回数（Acquisitions）を入力してください。
10

測定したデータは連番で保存されますが、保存時のファイル名を指定してください。
（Save Name）
Same Name を TEST 繰り返しを５回と指定した場合、データは TEST01,TEST02・・・TEST05の名前で保存されます。
Clear on each acquisition に☑をすると、繰り返し毎にデータを消去して測定します。

20

Figure 3 繰り返し測定の設定

4

KSPECT/MULTI SPECT 取り扱い説明書
2) Spectra タブ

Figure 4 Spctra タブa)カウント軸スケールの選択
カウント軸のスケールを選択します。Vertical Scale が Auto の場合、データの最大カウントを基準にして最適なスケールが自動的に設定されます。
10

Figure 5 カウント軸操作

5

KSPECT/MULTI SPECT 取り扱い説明書b)チャネル軸表示範囲の選択チャネル軸の拡大表示を選択します。拡大位置はカーソル位置を中心に拡大します。
Reset を選択することで、デフォルトのフルスケールに戻ります。

Figure 6 チャネル軸操作
拡大するとスペクトル表示が下図のように２分割された画面が表示されます。上段はフルスペクトルが10表示され、下段で表示している拡大エリアが網掛けで表示されます。上段の網掛け部をクリック＆ドラッグすることで下段の拡大表示位置を移動することができます。

Figure 7 拡大表示時の画面

6

KSPECT/MULTI SPECT 取り扱い説明書c)ROI 設定
スペクトルに ROI を設定するときに使用します。
ROI を設定したい領域の低エネルギー側にカーソルを移動して“Select Start”をクリックします。
次に ROI を設定したい領域の高エネルギー側にカーソルを移動して”Select End”をクリックするとスペクトル上に ROI が設定されます。
Show ROI をクリックすると設定した ROI の表示/非表示が切り替わります。
Subtract Background をクリックするとバックグラウンドエリアの表示/非表示が切り替わります。
10Clear はカーソルが置かれている ROI を消去するときに使用します。

Figure 8 ROI 設定

Figure 9 ROI 設定時の画面表示

7

KSPECT/MULTI SPECT 取り扱い説明書
ROI が設定されるとスペクトル情報エリアにカーソルが置かれた ROI の情報（Integral/Net カウント）が表示されます。

Figure 10 ROI 情報表示d) エネルギー表示モード（MSA/MSP のみ）
エネルギー較正されたスペクトルデータに対して X 軸のスケールをエネルギーとしてスペクトルを表示10します。エネルギー較正の異なる複数のスペクトルの比較に使用できます。
エネルギー表示モードでは、表示最大エネルギーを Range（Kev）にて 3Kev/2Kev/1Kev から選択できます。
Interval（Kev）はスペクトルドット数を選択できます。

Figure 11 エネルギー表示モード

2030

8

KSPECT/MULTI SPECT 取り扱い説明書
3) Analysis タブ （MSA/MSP のみ）
Analysis タブではエネルギー表示モードで以下の操作を実施できます。

Figure 12 Analysis
Σ（Create）：現在読み込まれているスペクトルの加算スペクトルを新規に作成します。加算対象のスペクトルは各々エネルギー較正されている必要があります。
10

Nuclide をクリックして開くポップアップ画面から分析対象としたい核種を選択することができます。

Figure 13 分析対象核種・エネルギーの選択

9

KSPECT/MULTI SPECT 取り扱い説明書

Figure 14 Analysis 画面
分析対象核種のピークがあればスペクトル上に核種名とエネルギー値が表示されます。
（Show ROI/Show Text で表示/非表示を切り替え可能）

Figure 15 分析結果画面
10

詳細設定・分析手法に関しては HELP（英文）を参照してください。

10

KSPECT/MULTI SPECT 取り扱い説明書

4.)Setup タブ
Setup タブでは検出器の LLD 設定とスペクトル情報の編集が実施できます。

Figure 16 Setup タブa) General タブ
10
General タブでは LLD の設定を変更することができます。（GR1 シリーズのみ）

Figure 17 General タブ
Set LLD をクリックすると赤色にボタンが変わるとともに、スペクトル上に現在設定されている LLD 位置が赤いカーソールで表示されます。
新しく設定した LLD 位置に白いカーソルを合わせて SetLLD をクリックするとボタンが白に切り替
わり LLD が設定されます。
20b) エネルギー較正タブ

11

KSPECT/MULTI SPECT 取り扱い説明書
エネルギー較正を設定します。
（P.3 ２-1）-ｄ）を参照ください）c) エネルギー較正タブ(MSA/MSP のみ)
半値幅較正を設定します。ピークエネルギーとそのエネルギーピークでの FWHM を登録します。

10
Figure 18 半値幅較正画面d) 効率較正タブ(MSP のみ)
効率較正を実施します。
効率較正はピークエネルギーとそのエネルギーにおける効率を設定します。

12

KSPECT/MULTI SPECT 取り扱い説明書

Figure 19 効率較正画面
効率計算はピーク核種とエネルギーを指定後、効率較正で使用した線源情報を入力します。

Figure 20 効率計算画面

1013

KSPECT/MULTI SPECT 取り扱い説明書e) 測定情報の表示
Measurement Information から測定データの情報を確認することができます。
MSA/MSP では情報の変更が可能です。

1014

KSPECT/MULTI SPECT 取り扱い説明書f)

制御対象の選択（KSpect のみ）
MSA/MSP は同時に複数の KromeK 社検出器・MCA を制御することができますが、KSpect は１つのデバイスのみ制御が可能になっています。
PC に複数の KromeK 製品を接続した場合、KSpect で制御する対象を１つ選択する必要があります。

Figure 21 制御デバイスの選択
10

ここで選択されたデバイスの制御を実施します。

5) Report タブ
Report タブでは KSpect/MSA/MSP に応じた帳票（英文）を出力します。
各ブログラムごとに帳票に出力する項目を選択できます。

20

Figure 22 帳票項目の選択（MSA の場合）

15

KSPECT/MULTI SPECT 取り扱い説明書

6.File タブ

File をクリックするとプルダウンメニューが表示されます。

Figure 23 File メニュー
1) Measurement Database10

KSpect/MSA/MSP で保存したデータは Microsoft 社の SQL データベースに登録されます。

Figure 24 データベース

16

KSPECT/MULTI SPECT 取り扱い説明書

Save で保存したデータはいったんデータベースに格納されます。
測定データを個別のファイルとして保存する場合には保存したファイルをクリックで選択してExport を実行してください。

B.メイン画面
メイン画面には測定中のスペクトルがリアルタイムで表示されます。また、保存したデータを読込んで表示します。
10

1. Spectra タブ

Figure 25 スペクトル表示画面

17

KSPECT/MULTI SPECT 取り扱い説明書

2. Monitor タブ (MSA/MSP のみ)
Monitor タブでは全スペクトルや任意の ch 幅（エネルギー幅）の積算カウントのトレンドグラフを選択した時間単位で更新することができます。

Figure 26 Monitor 画面

1020

18

KSPECT/MULTI SPECT 取り扱い説明書

C.情報画面

情報表示画面には測定中データに関する情報が表示されます。

Figure 27 情報表示画面

10Real time

: 測定時間（実時間）

Live time: 実測定時間

Dead time: 不感率Count rate

: 全カウントの cps

Scale ranges
Vertical: 表示しているカウント軸の最大値

Horizontal: 表示しているｃｈ軸の最大値
エネルギー較正が実施されている場合にはエネルギー値も表示

Cursor data
20

Vertical: カーソル位置のカウント数

Horizontal: カーソル位置のｃｈ番
エネルギー較正が実施されている場合にはエネルギー値も表示

ROI data
Integral: ROI が設定されている場合で、ROI 領域内にカーソルがあるとき、その ROI の全計数値を表示

Net: ROI が設定されている場合で、ROI 領域内にカーソルがあるとき、その ROI のネットカウント値を表示

Centroid: ROI が設定されている場合で、ROI 領域内にカーソルがあるとき、その ROI の中心ｃｈを表示

30

エネルギー較正が実施されている場合にはエネルギー値も表示
FWHM：ROI が設定されている場合で、ROI 領域内にカーソルがあるとき、その ROI のFWHM を表示
エネルギー較正が実施されている場合にはエネルギー値も表示

Active device

19

KSPECT/MULTI SPECT 取り扱い説明書
Type: 対象データの検出・MCA タイプ

S/N: 対象データの検出器・MCA の製造番号

LLD: LLD 設定値

SCO

: この表示は未使用です。

D ステータス画面
PC と検出器・MCA が正常に通信しているとき、ステータス画面には Fig28 のステータス画面が表示されます。
10Figure 28 ステータス画面

2020

Appendix 1 スペクトルデータフォーマット
概要

スペクトルを個別ファイルとして保存した場合のデータフォーマット例。データはすべて ASCII で保存されます。選択できるフォーマットはオリジナル
フォーマットの SPE、ANSI フォーマットの.n42 (XML),及び .txt が選択できます。

.spe$SPEC_REM:
Multi-Spect
ID:
NAME: 137Cs GR-1

1200sec

I:\Spectrum_Point\GR-1\137Cs$DATE_MEA:
12/04/2014 15:37:39$MEAS_TIM:
1200.005 1201.575$DATA:
0 4095000000000000000000

GR-1

1200sec.spe

.n42<?xml version="1.0" encoding="utf-8"?><RadInstrumentData

xmlns:xsi="http://www.w3.org/2001/XMLSchema-instance"

xsi:schemaLocation="http://physics.nist.gov/N42/2011/N42
http://physics.nist.gov/N42/2011/n42.xsd"
25773279e9e9"n42DocUUID="769f2f6a-33ab-4bd5-a603-

n42DocDateTime="2018-08-30T14:00:06+09:00"

xmlns="http://physics.nist.gov/N42/2011/N42"><Remark>Exportedfrom

KromekMultiSpect

Software-http://www.kromek.com</Remark><RadInstrumentDataCreatorName>HPDESKTOP\jsait</RadInstrumentDataCreatorName><RadInstrumentInformation id="HP-DESKTOP"><RadInstrumentManufacturerName>Kromek
Ltd</RadInstrumentManufacturerName><RadInstrumentIdentifier>3189</RadInstrumentIdentifier><RadInstrumentModelName>MultiSpect/KSpect</RadInstrumentModelName><RadInstrumentClassCode>Other</RadInstrumentClassCode><RadInstrumentVersion><RadInstrumentComponentName>Software
Version</RadInstrumentComponentName><RadInstrumentComponentVersion>Kromekv14.18.4.201</RadInstrumentComponentVersion></RadInstrumentVersion></RadInstrumentInformation><RadDetectorInformation id="Instrument_3189"><RadDetectorName>3189</RadDetectorName><RadDetectorCategoryCode>Gamma</RadDetectorCategoryCode><RadDetectorKindCode>CZT</RadDetectorKindCode><RadDetectorDescription>GR1A</RadDetectorDescription><RadDetectorLengthValue units="cm">1</RadDetectorLengthValue>

KSpect<RadDetectorWidthValue units="cm">1</RadDetectorWidthValue><RadDetectorDepthValue units="cm">1</RadDetectorDepthValue><RadDetectorCharacteristics><Characteristic><CharacteristicName>LLD</CharacteristicName><CharacteristicValue>32</CharacteristicValue><CharacteristicValueUnits>channels</CharacteristicValueUnits><CharacteristicValueDataClassCode>int</CharacteristicValueDataClassCode></Characteristic><Characteristic><CharacteristicName>SCO</CharacteristicName><CharacteristicValue>-2147483648</CharacteristicValue><CharacteristicValueUnits>channels</CharacteristicValueUnits><CharacteristicValueDataClassCode>int</CharacteristicValueDataClassCode></Characteristic><Characteristic><CharacteristicName>DEAD_TIME_FACTOR</CharacteristicName><CharacteristicValue>1E-05</CharacteristicValue><CharacteristicValueUnits>cps</CharacteristicValueUnits><CharacteristicValueDataClassCode>double</CharacteristicValueDataClassCode></Characteristic><Characteristic><CharacteristicName>DETECTOR_TYPE</CharacteristicName><CharacteristicValue>257</CharacteristicValue><CharacteristicValueUnits>Not Applicable</CharacteristicValueUnits><CharacteristicValueDataClassCode>int</CharacteristicValueDataClassCode></Characteristic><Characteristic><CharacteristicName>PRODUCT_FAMILY</CharacteristicName><CharacteristicValue>PC</CharacteristicValue><CharacteristicValueUnits>Not Applicable</CharacteristicValueUnits><CharacteristicValueDataClassCode>int</CharacteristicValueDataClassCode></Characteristic></RadDetectorCharacteristics></RadDetectorInformation>

<RadMeasurement id="Run6_1"><MeasurementClassCode>Foreground</MeasurementClassCode><StartDateTime>2018-08-30T14:00:06+09:00</StartDateTime><RealTimeDuration>PT1261.001S</RealTimeDuration><Spectrum

id="Spectrum_Run6_1"

energyCalibrationReference="EnergyCalibration_Run6_1"
radDetectorInformationReference="Instrument_3189"
FWHMCalibrationReference="ResolutionCalibration_Run6_1"><LiveTimeDuration>PT1258.205S</LiveTimeDuration><ChannelData>0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 1 27 70
218 478 797 </ChannelData></Spectrum></RadMeasurement><RadMeasurement id="Background_Run6_1"><MeasurementClassCode>Background</MeasurementClassCode><StartDateTime>2018-08-30T14:00:06+09:00</StartDateTime><RealTimeDuration>PT1261.001S</RealTimeDuration><Spectrum

id="Spectrum_Back_Run6_1"

energyCalibrationReference="EnergyCalibration_Run6_1"
radDetectorInformationReference="Instrument_3189"
FWHMCalibrationReference="ResolutionCalibration_Run6_1"><LiveTimeDuration>PT0S</LiveTimeDuration><ChannelData>0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 0 00 0 0 0 0 0 0 </ChannelData></Spectrum></RadMeasurement><EnergyCalibration id="EnergyCalibration_Run6_1"><CoefficientValues>-1.76382371062346

0.758668925588912

0.0</CoefficientValues><CalibrationDateTime>2018-08-30T14:22:18+09:00</CalibrationDateTime></EnergyCalibration><FWHMCalibration id="ResolutionCalibration_Run6_1"><EnergyValues>59.5409 122.06065 661.657 </EnergyValues><FWHMValues>8.0380215 7.323639 16.541425 </FWHMValues></FWHMCalibration></RadInstrumentData>

.txt
00000000000000000000000000000001
2770
218

478
797
1176
1543
1817
1999
2284
2448
2860
2989
3287
3164
3519
3230
3186
2987
2670
2408
2165
1857
1610
1216
1060942