BA01253D/06/EN/03.23-00
71601225
2023-02-01

Products

Solutions

Valid as of version
01.01.zz (Device firmware)

Operating Instructions
Proline Promass P 100
PROFIBUS DP
Coriolis flowmeter

Services

Proline Promass P 100 PROFIBUS DP• Make sure the document is stored in a safe place such that it is always available whenworking on or with the device.
• To avoid danger to individuals or the facility, read the "Basic safety instructions" section
carefully, as well as all other safety instructions in the document that are specific toworking procedures.
• The manufacturer reserves the right to modify technical data without prior notice. Your
Endress+Hauser Sales Center will supply you with current information and updates tothese instructions.

2

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Table of contents

Table of contents
1

About this document . . . . . . . . . . . . . . . . 6

1.1
1.2

1.4

Document function . . . . . . . . . . . . . . . . . . . . . 6
Symbols . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6
1.2.1 Safety symbols . . . . . . . . . . . . . . . . . . 6
1.2.2 Electrical symbols . . . . . . . . . . . . . . . . 6
1.2.3 Tool symbols . . . . . . . . . . . . . . . . . . . . 6
1.2.4 Symbols forcertain types of information . . . . . . . . . 7
1.2.5 Symbols in graphics . . . . . . . . . . . . . . . 7
Documentation . . . . . . . . . . . . . . . . . . . . . . . . 7
1.3.1 Document function . . . . . . . . . . . . . . . 8
Registered trademarks . . . . . . . . . . . . . . . . . . . 82

Safety instructions . . . . . . . . . . . . . . . . . . 9

2.1
2.2
2.3
2.4
2.5
2.6

Requirements for the personnel . . . . . . . . . . . . 9
Intended use . . . . . . . . . . . . . . . . . . . . . . . . . . 9
Workplace safety . . . . . . . . . . . . . . . . . . . . . . 10
Operational safety . . . . . . . . . . . . . . . . . . . . . 10
Product safety . . . . . . . . . . . . . . . . . . . . . . . . 10
IT security . . . . . . . . . . . . . . . . . . . . . . . . . . . 10

3

Product description . . . . . . . . . . . . . . . . 11

3.1

Product design . . . . . . . . . . . . . . . . . . . . . . . . 11
3.1.1 Device version with PROFIBUS DP
communication protocol . . . . . . . . . . 11

1.3

4

Incoming acceptance and productidentification . . . . . . . . . . . . . . . . . . . . . 12

4.1
4.2

Incoming acceptance . . . . . . . . . . . . . . . . . . . 12
Product identification . . . . . . . . . . . . . . . . . . . 12
4.2.1 Transmitter nameplate . . . . . . . . . . . 13
4.2.2 Sensor nameplate . . . . . . . . . . . . . . . 14
4.2.3 Symbols on measuring device . . . . . . 15

5

Storage and transport . . . . . . . . . . . . . 16

5.1
5.2

5.3

Storage conditions . . . . . . . . . . . . . . . . . . . . . 16
Transporting the product . . . . . . . . . . . . . . . . 16
5.2.1 Measuring devices without lifting
lugs . . . . . . . . . . . . . . . . . . . . . . . . . 16
5.2.2 Measuring devices with lifting lugs . . 17
5.2.3 Transporting with a fork lift . . . . . . . . 17
Packaging disposal . . . . . . . . . . . . . . . . . . . . . 17

6

Mounting . . . . . . . . . . . . . . . . . . . . . . . . . 18

6.1

Mounting requirements . . . . . . . . . . . . . . . . . 18
6.1.1 Mounting position . . . . . . . . . . . . . . . 18
6.1.2 Environment and process
requirements . . . . . . . . . . . . . . . . . . 20
6.1.3 Special mounting instructions . . . . . . 22
Mounting the measuring device . . . . . . . . . . . 23
6.2.1 Required tool . . . . . . . . . . . . . . . . . . 23

6.2

Endress+Hauser

6.3

6.2.2 Preparing the measuring device . . . . . 23
6.2.3 Mounting the measuring device . . . . . 23
6.2.4 Turning the display module . . . . . . . . 24
Post-installation check . . . . . . . . . . . . . . . . . . 25

7

Electrical connection . . . . . . . . . . . . . . 26

7.1
7.2

7.6
7.7

Electrical safety . . . . . . . . . . . . . . . . . . . . . . . 26
Connecting requirements . . . . . . . . . . . . . . . . 26
7.2.1 Required tools . . . . . . . . . . . . . . . . . . 26
7.2.2 Requirements for connecting cable . . . 26
7.2.3 Terminal assignment . . . . . . . . . . . . . 27
7.2.4 Pin assignment, device plug . . . . . . . . 28
7.2.5 Preparing the measuring device . . . . . 29
Connecting the measuring device . . . . . . . . . . 29
7.3.1 Connecting the transmitter . . . . . . . . 29
7.3.2 Potential equalization . . . . . . . . . . . . 31
Special connection instructions . . . . . . . . . . . . 31
7.4.1 Connection examples . . . . . . . . . . . . . 31
Hardware settings . . . . . . . . . . . . . . . . . . . . . 31
7.5.1 Setting the device address . . . . . . . . . 31
7.5.2 Enabling the terminating resistor . . . . 32
Ensuring the degree of protection . . . . . . . . . . 33
Post-connection check . . . . . . . . . . . . . . . . . . 33

8Operation options . . . . . . . . . . . . . . . . . 35

8.1
8.2

Overview of operating options . . . . . . . . . . . . 35
Structure and function of the operating
menu . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 36
8.2.1 Structure of the operating menu . . . . 36
8.2.2 Operating philosophy . . . . . . . . . . . . 37
Displaying the measured values via the local
display (optionally available) . . . . . . . . . . . . . 38
8.3.1 Operational display . . . . . . . . . . . . . . 38
8.3.2 User roles and related accessauthorization . . . . . . . . . . . . . . . . . . 39
Access to operating menu via web browser . . . 40
8.4.1 PROFINET with Ethernet-APL . . . . . . 40
8.4.2 Prerequisites . . . . . . . . . . . . . . . . . . . 40
8.4.3 Establishing a connection . . . . . . . . . 41
8.4.4 Logging on . . . . . . . . . . . . . . . . . . . . 42
8.4.5 User interface . . . . . . . . . . . . . . . . . . 43
8.4.6 Disabling the Web server . . . . . . . . . . 44
8.4.7 Logging out . . . . . . . . . . . . . . . . . . . . 45
Access to the operating menu via theoperating tool . . . . . . . . . . . . . . . . . . . . . . . . 45
8.5.1 Connecting the operating tool . . . . . . 45
8.5.2 FieldCare . . . . . . . . . . . . . . . . . . . . . 46
8.5.3 DeviceCare . . . . . . . . . . . . . . . . . . . . 47

7.3
7.4
7.5

8.3

8.4

8.5

9

System integration . . . . . . . . . . . . . . . . 48

9.1

Overview of device description files . . . . . . . . . 48
9.1.1 Current version data for the device . . . 48
9.1.2 Operating tools . . . . . . . . . . . . . . . . . 48

3

Table of contents

9.2
9.3

9.4

10

Device master file (GSD) . . . . . . . . . . . . . . . . . 48
9.2.1 Manufacturer-specific GSD . . . . . . . . 49
9.2.2 Profile GSD . . . . . . . . . . . . . . . . . . . . 49
Integration into a PROFIBUS network . . . . . . . 50
9.3.1 Block model . . . . . . . . . . . . . . . . . . . 50
9.3.2 Assignment of the measured values
in the function blocks . . . . . . . . . . . . 50
9.3.3 Totalizer control SET_TOT . . . . . . . . . 51
Cyclic data transmission . . . . . . . . . . . . . . . . 52
9.4.1 Block model . . . . . . . . . . . . . . . . . . . 52
9.4.2 Description of the modules . . . . . . . . 52

Commissioning . . . . . . . . . . . . . . . . . . . . 58

10.1 Post-installation and post-connection check . . 58
10.2 Connecting via FieldCare . . . . . . . . . . . . . . . . 58
10.3 Setting the operating language . . . . . . . . . . . . 58
10.4 Configuring the measuring device . . . . . . . . . . 58
10.4.1 Defining the tag name . . . . . . . . . . . . 59
10.4.2 Setting the system units . . . . . . . . . . 59
10.4.3 Selecting and setting the medium . . . 62
10.4.4 Configuring communicationinterface . . . . . . . . . . . . . . . . . . . . . . 63
10.4.5 Configuration of the Analog Inputs . . 64
10.4.6 Configuring the low flow cut off . . . . . 65
10.4.7 Configuring partially filled pipe
detection . . . . . . . . . . . . . . . . . . . . . 66
10.5 Advanced settings . . . . . . . . . . . . . . . . . . . . . 67
10.5.1 Using the parameter to enter theaccess code . . . . . . . . . . . . . . . . . . . . 67
10.5.2 Calculated process variables . . . . . . . . 67
10.5.3 Carrying out a sensor adjustment . . . . 69
10.5.4 Configuring the totalizer . . . . . . . . . . 73
10.5.5 Using parameters for device
administration . . . . . . . . . . . . . . . . . 74
10.6 Simulation . . . . . . . . . . . . . . . . . . . . . . . . . . . 75
10.7 Protecting settings from unauthorized access . 76
10.7.1 Write protection via access code . . . . . 76
10.7.2 Write protection via write protection
switch . . . . . . . . . . . . . . . . . . . . . . . . 77

11

Operation . . . . . . . . . . . . . . . . . . . . . . . . . 78

11.1 Reading the device locking status . . . . . . . . . . 78
11.2 Adjusting the operating language . . . . . . . . . . 78
11.3 Configuring the display . . . . . . . . . . . . . . . . . 78
11.4 Reading measured values . . . . . . . . . . . . . . . . 78
11.4.1 "Measured variables" submenu . . . . . . 78
11.4.2 "Totalizer" submenu . . . . . . . . . . . . . . 80
11.5 Adapting the measuring device to the process
conditions . . . . . . . . . . . . . . . . . . . . . . . . . . . 81
11.6 Performing a totalizer reset . . . . . . . . . . . . . . 81

12

Diagnostics and troubleshooting . . . 83

12.1
12.2

General troubleshooting . . . . . . . . . . . . . . . . . 83
Diagnostic information via light emitting
diodes . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 84
12.2.1 Transmitter . . . . . . . . . . . . . . . . . . . . 84

4

Proline Promass P 100 PROFIBUS DP

12.3

Diagnostic information in the Web browser . . 85
12.3.1 Diagnostic options . . . . . . . . . . . . . . . 85
12.3.2 Calling up remedy information . . . . . . 86
12.4 Diagnostic information in FieldCare or
DeviceCare . . . . . . . . . . . . . . . . . . . . . . . . . . 86
12.4.1 Diagnostic options . . . . . . . . . . . . . . . 86
12.4.2 Calling up remedy information . . . . . . 87
12.5 Adapting the diagnostic information . . . . . . . 87
12.5.1 Adapting the diagnostic behavior . . . . 87
12.6 Overview of diagnostic information . . . . . . . . 90
12.6.1 Diagnostic of sensor . . . . . . . . . . . . . 90
12.6.2 Diagnostic of electronic . . . . . . . . . . . 94
12.6.3 Diagnostic of configuration . . . . . . . 102
12.6.4 Diagnostic of process . . . . . . . . . . . . 108
12.7 Pending diagnostic events . . . . . . . . . . . . . . 115
12.8 Diagnostic list . . . . . . . . . . . . . . . . . . . . . . . 116
12.9 Event logbook . . . . . . . . . . . . . . . . . . . . . . . 116
12.9.1 Reading out the event logbook . . . . . 116
12.9.2 Filtering the event logbook . . . . . . . 117
12.9.3 Overview of information events . . . . 117
12.10 Resetting the measuring device . . . . . . . . . . 118
12.10.1 Function scope of "Device reset"
parameter . . . . . . . . . . . . . . . . . . . . 118
12.11 Device information . . . . . . . . . . . . . . . . . . . 119
12.12 Firmware history . . . . . . . . . . . . . . . . . . . . . 121

13

Maintenance . . . . . . . . . . . . . . . . . . . . 122

13.1
13.2
13.3

Maintenance tasks . . . . . . . . . . . . . . . . . . . . 122
13.1.1 Exterior cleaning . . . . . . . . . . . . . . . 122
13.1.2 Interior cleaning . . . . . . . . . . . . . . . 122
Measuring and test equipment . . . . . . . . . . . 122
Endress+Hauser services . . . . . . . . . . . . . . . 122

14

Repair . . . . . . . . . . . . . . . . . . . . . . . . . . . 123

14.1

General information . . . . . . . . . . . . . . . . . . 123
14.1.1 Repair and conversion concept . . . . . 123
14.1.2 Notes for repair and conversion . . . . 123
14.2 Spare parts . . . . . . . . . . . . . . . . . . . . . . . . . 123
14.3 Endress+Hauser services . . . . . . . . . . . . . . . 123
14.4 Return . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 123
14.5 Disposal . . . . . . . . . . . . . . . . . . . . . . . . . . . 124
14.5.1 Removing the measuring device . . . . 124
14.5.2 Disposing of the measuring device . . 124

15

Accessories . . . . . . . . . . . . . . . . . . . . . . 125

15.1
15.2
15.3
15.4

Device-specific accessories . . . . . . . . . . . . . . 125
15.1.1 For the sensor . . . . . . . . . . . . . . . . . 125
Communication-specific accessories . . . . . . . 125
Service-specific accessories . . . . . . . . . . . . . . 126
System components . . . . . . . . . . . . . . . . . . . 127

16

Technical data . . . . . . . . . . . . . . . . . . . 128

16.1
16.2
16.3
16.4

Application . . . . . . . . . . . . . . . . . . . . . . . . . 128
Function and system design . . . . . . . . . . . . . 128
Input . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 129
Output . . . . . . . . . . . . . . . . . . . . . . . . . . . . 130
Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Table of contents

16.5 Power supply . . . . . . . . . . . . . . . . . . . . . . . . 132
16.6 Performance characteristics . . . . . . . . . . . . . 133
16.7 Mounting . . . . . . . . . . . . . . . . . . . . . . . . . . 136
16.8 Environment . . . . . . . . . . . . . . . . . . . . . . . . 136
16.9 Process . . . . . . . . . . . . . . . . . . . . . . . . . . . . 137
16.10 Mechanical construction . . . . . . . . . . . . . . . 140
16.11 Operability . . . . . . . . . . . . . . . . . . . . . . . . . 143
16.12 Certificates and approvals . . . . . . . . . . . . . . 145
16.13 Application packages . . . . . . . . . . . . . . . . . . 147
16.14 Accessories . . . . . . . . . . . . . . . . . . . . . . . . . 148
16.15 Supplementary documentation . . . . . . . . . . . 148

Index . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 150

Endress+Hauser

5About this document

Proline Promass P 100 PROFIBUS DP

1

About this document

1.1

Document function

These Operating Instructions contain all the information required in the various life cycle
phases of the device: from product identification, incoming acceptance and storage, toinstallation, connection, operation and commissioning, through to troubleshooting,
maintenance and disposal.

1.2

Symbols

1.2.1

Safety symbols

DANGER

This symbol alerts you to a dangerous situation. Failure to avoid this situation will result inserious or fatal injury.
WARNING

This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inserious or fatal injury.
CAUTION

This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inminor or medium injury.
NOTICE

This symbol contains information on procedures and other facts which do not result inpersonal injury.

1.2.2

Electrical symbols
SymbolMeaning
Direct current
Alternating current
Direct current and alternating current
Ground connection
A grounded terminal which, as far as the operator is concerned, is grounded via agrounding system.
Potential equalization connection (PE: protective earth)Ground terminals that must be connected to ground prior to establishing any otherconnections.
The ground terminals are located on the interior and exterior of the device:• Interior ground terminal: potential equalization is connected to the supply network.
• Exterior ground terminal: device is connected to the plant grounding system.

1.2.3

Tool symbols
SymbolMeaning
Allen key
Open-ended wrench

6

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

About this document

1.2.4

Symbols for certain types of information
SymbolMeaning
Permitted
Procedures, processes or actions that are permitted.
Preferred
Procedures, processes or actions that are preferred.
Forbidden
Procedures, processes or actions that are forbidden.
Tip
Indicates additional information.
Reference to documentation
Reference to pageA

Reference to graphic
Notice or individual step to be observed
1. , 2. , 3. …

Series of steps
Result of a step
Help in the event of a problem
Visual inspection

1.2.5

Symbols in graphics
SymbolMeaning

1, 2, 3, ...

Item numbers

1. , 2. , 3. , …

Series of steps

A, B, C, ...

Views

A-A, B-B, C-C, ...

Sections-

Hazardous area.

Safe area (non-hazardous area)
Flow direction

1.3

Documentation
For an overview of the scope of the associated Technical Documentation, refer to thefollowing:• Device Viewer (www.endress.com/deviceviewer): Enter the serial number from thenameplate• Endress+Hauser Operations app: Enter serial number from nameplate or scanmatrix code on nameplate.

Endress+Hauser

7About this document

Proline Promass P 100 PROFIBUS DP

1.3.1

Document function

The following documentation may be available depending on the version ordered:
Document type

Purpose and content of the document

Technical Information (TI)

Planning aid for your device
The document contains all the technical data on the device and provides
an overview of the accessories and other products that can be ordered forthe device.

Brief Operating Instructions (KA)

Guide that takes you quickly to the 1st measured value
The Brief Operating Instructions contain all the essential informationfrom incoming acceptance to initial commissioning.

Operating Instructions (BA)Your reference document
The Operating Instructions contain all the information that is required in
the various phases of the life cycle of the device: from productidentification, incoming acceptance and storage, to mounting,connection, operation and commissioning through to troubleshooting,
maintenance and disposal.

Description of Device Parameters(GP)

Reference for your parameters
The document provides a detailed explanation of each individual
parameter. The description is aimed at those who work with the device
over the entire life cycle and perform specific configurations.

Safety Instructions (XA)

Depending on the approval, safety instructions for electrical equipment inhazardous areas are also supplied with the device. The Safety Instructions
are an integral part of the Operating Instructions.
Information on the Safety Instructions (XA) relevant to the device is provided
on the nameplate.

Supplementary device-dependent
documentation (SD/FY)

1.4

Always comply strictly with the instructions in the relevant
supplementary documentation. The supplementary documentation is an
integral part of the device documentation.

Registered trademarks

PROFIBUS®
Registered trademark of the PROFIBUS Nutzerorganisation e.V. (PROFIBUS User
Organization), Karlsruhe, Germany
TRI-CLAMP®
Registered trademark of Ladish & Co., Inc., Kenosha, USA

8

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Safety instructions

2

Safety instructions

2.1

Requirements for the personnel

The personnel for installation, commissioning, diagnostics and maintenance must fulfill
the following requirements:‣ Trained, qualified specialists must have a relevant qualification for this specific function
and task.
‣ Are authorized by the plant owner/operator.
‣ Are familiar with federal/national regulations.
‣ Before starting work, read and understand the instructions in the manual andsupplementary documentation as well as the certificates (depending on theapplication).
‣ Follow instructions and comply with basic conditions.
The operating personnel must fulfill the following requirements:‣ Are instructed and authorized according to the requirements of the task by the facility'sowner-operator.

‣ Follow the instructions in this manual.

2.2

Intended use

Application and media
The measuring device described in this manual is intended only for the flow measurementof liquids.
Depending on the version ordered, the measuring device can also measure potentiallyexplosive, flammable, poisonous and oxidizing media.
Measuring devices for use in hazardous areas, in hygienic applications or where there is anincreased risk due to process pressure, are labeled accordingly on the nameplate.
To ensure that the measuring device remains in proper condition for the operation time:‣ Keep within the specified pressure and temperature range.
‣ Only use the measuring device in full compliance with the data on the nameplate andthe general conditions listed in the Operating Instructions and supplementary
documentation.
‣ Based on the nameplate, check whether the ordered device is permitted for theintended use in the hazardous area (e.g. explosion protection, pressure vessel safety).
‣ Use the measuring device only for media to which the process-wetted materials are
sufficiently resistant.
‣ If the ambient temperature of the measuring device is outside the atmospherictemperature, it is absolutely essential to comply with the relevant basic conditions as
specified in the device documentation →  7.
‣ Protect the measuring device permanently against corrosion from environmental
influences.
Incorrect use
Non-designated use can compromise safety. The manufacturer is not liable for damage
caused by improper or non-intended use.

LWARNING
Danger of breakage due to corrosive or abrasive fluids and ambient conditions!
‣ Verify the compatibility of the process fluid with the sensor material.
‣ Ensure the resistance of all fluid-wetted materials in the process.
‣ Keep within the specified pressure and temperature range.

Endress+Hauser

9Safety instructions

Proline Promass P 100 PROFIBUS DPNOTICE
Verification for borderline cases:‣ For special fluids and fluids for cleaning, Endress+Hauser is glad to provide assistancein verifying the corrosion resistance of fluid-wetted materials, but does not accept any
warranty or liability as minute changes in the temperature, concentration or level ofcontamination in the process can alter the corrosion resistance properties.
Residual risks

LCAUTION
The electronics and the medium may cause the surfaces to heat up or freeze. Risk ofburns or frostbite!
‣ Mount suitable touch protection.

2.3

Workplace safety

When working on and with the device:‣ Wear the required personal protective equipment as per national regulations.

2.4

Operational safety

Damage to the device!

‣ Operate the device in proper technical condition and fail-safe condition only.
‣ The operator is responsible for the interference-free operation of the device.
Modifications to the device
Unauthorized modifications to the device are not permitted and can lead to unforeseeable
dangers!
‣ If modifications are nevertheless required, consult with the manufacturer.
RepairTo ensure continued operational safety and reliability:‣ Carry out repairs on the device only if they are expressly permitted.
‣ Observe federal/national regulations pertaining to the repair of an electrical device.
‣ Use only original spare parts and accessories.

2.5

Product safety

This measuring device is designed in accordance with good engineering practice to meetstate-of-the-art safety requirements, has been tested, and left the factory in a condition inwhich it is safe to operate.
It meets general safety standards and legal requirements. It also complies with the EU
directives listed in the device-specific EU Declaration of Conformity. The manufacturer
confirms this by affixing the CE mark to the device..

2.6

IT security

Our warranty is valid only if the product is installed and used as described in the Operating
Instructions. The product is equipped with security mechanisms to protect it against any
inadvertent changes to the settings.
IT security measures, which provide additional protection for the product and associateddata transfer, must be implemented by the operators themselves in line with their security
standards.

10

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

3

Product description

Product description

The device consists of a transmitter and a sensor.
The device is available as a compact version:
The transmitter and sensor form a mechanical unit.

3.1

Product design

3.1.1

Device version with PROFIBUS DP communication protocol

5

4

6

7
3

2

1

A0023153

1
1
2
3
4
5
6
7

Endress+Hauser

Important components of a measuring device
SensorTransmitter housing
Main electronics module
Transmitter housing coverTransmitter housing cover (version for optional local display)
Local display (optional)
Main electronics module (with bracket for optional local display)11

Incoming acceptance and product identification

Proline Promass P 100 PROFIBUS DP

4

Incoming acceptance and productidentification

4.1

Incoming acceptance
Are the order codes on thedelivery note (1) and the
product sticker (2) identical?

1
2

1
2

Are the goods undamaged?

Do the data on the
nameplate match theordering information on thedelivery note?

Order code:
Ser. no.:Ext. ord. cd.:

i

i

Date:

Is the envelope present withaccompanying documents?

• If one of the conditions is not satisfied, contact your Endress+Hauser Sales Center.
• The Technical Documentation is available via the Internet or via the Endress+Hauser
Operations App, see the "Product identification" section →  13.

4.2

Product identification

The following options are available for identification of the device:• Nameplate specifications• Order code with breakdown of the device features on the delivery note
• Enter the serial numbers from the nameplates in the Device Viewer(www.endress.com/deviceviewer): all the information about the device is displayed.
• Enter the serial numbers from the nameplates into the Endress+Hauser Operations App
or scan the DataMatrix code on the nameplate with the Endress+Hauser Operations App:
all the information about the device is displayed.

12

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Incoming acceptance and product identification

For an overview of the scope of the associated Technical Documentation, refer to thefollowing:• The chapters "Additional standard documentation on the device" and "Supplementarydevice-dependent documentation"• The Device Viewer: Enter the serial number from the nameplate(www.endress.com/deviceviewer)• The Endress+Hauser Operations App: Enter the serial number from the nameplate orscan the DataMatrix code on the nameplate.

4.2.1

Transmitter nameplate
7
1
2
3
4
5

Order code:
Ser. no.:Ext. ord. cd.:

8
i

6

9
13

12

11

10
A0030222

2
1
2
3
4
5
6
7
8
9
10
11
12
13

Endress+Hauser

Example of a transmitter nameplate
Manufacturing location
Name of the transmitter
Order code
Serial number (Ser. no.)Extended order code (Ext. ord. cd.)
Electrical connection data, e.g. available inputs and outputs, supply voltage
Permitted ambient temperature (Ta)
Degree of protection
2-D matrix code
Document number of safety-related supplementary documentation →  148
Date of manufacture: year-month
CE mark, RCM-Tick markFirmware version (FW)13

Incoming acceptance and product identification

4.2.2

Proline Promass P 100 PROFIBUS DP

Sensor nameplate
1

2
3
4
5

Order code:
Ser. no.:Ext. ord. cd.:

6

7
14
13
i

8

i
Date:

9
10

12

11
A0029199

3
1
2
3
4
5
6

7
8
9
10
11
12
13
14

Example of a sensor nameplate
Name of the sensor
Place of manufacture
Order code
Serial number (Ser. no.)Extended order code (Ext. ord. cd.)
Nominal diameter of the sensor; flange nominal diameter/nominal pressure; sensor test pressure; medium
temperature range; material of measuring tube and manifold; sensor-specific information: e.g. pressure rangeof sensor housing, wide-range density specification (special density calibration)
Approval information for explosion protection, Pressure Equipment Directive and degree of protection
Flow direction
Manufacturing date: year-month2-D matrix code
Document number of safety-related supplementary documentation
CE mark, RCM-Tick markSurface roughness
Permitted ambient temperature (Ta)

Order code
The measuring device is reordered using the order code.
Extended order code
• The device type (product root) and basic specifications (mandatory features) arealways listed.
• Of the optional specifications (optional features), only the safety and approvalrelated specifications are listed (e.g. LA). If other optional specifications are alsoordered, these are indicated collectively using the # placeholder symbol (e.g. #LA#).
• If the ordered optional specifications do not include any safety and approval-related
specifications, they are indicated by the + placeholder symbol (e.g. XXXXXX-ABCDE+).

14

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

4.2.3
SymbolIncoming acceptance and product identification

Symbols on measuring device
MeaningWARNING!
This symbol alerts you to a dangerous situation. Failure to avoid this situation can result in serious
or fatal injury. To determine the nature of the potential hazard and the measures required to
avoid it, consult the documentation accompanying the measuring device.
Reference to documentation
Refers to the corresponding device documentation.
Protective ground connection
A terminal which must be connected to ground prior to establishing any other connections.

Endress+Hauser

15

Storage and transport

Proline Promass P 100 PROFIBUS DP

5Storage and transport

5.1

Storage conditions

Observe the following notes for storage:‣ Store in the original packaging to ensure protection from shock.
‣ Do not remove protective covers or protective caps installed on process connections.
They prevent mechanical damage to the sealing surfaces and contamination in themeasuring pipe.
‣ Protect from direct sunlight to avoid unacceptably high surface temperatures.
‣ Store in a dry and dust-free place.
‣ Do not store outdoors.
Storage temperature →  137

5.2

Transporting the productTransport the measuring device to the measuring point in the original packaging.

A0029252

Do not remove protective covers or caps installed on process connections. They
prevent mechanical damage to the sealing surfaces and contamination in themeasuring tube.

5.2.1

Measuring devices without lifting lugs

LWARNING
Center of gravity of the measuring device is higher than the suspension points of the
webbing slings.
Risk of injury if the measuring device slips.
‣ Secure the measuring device against slipping or turning.
‣ Observe the weight specified on the packaging (stick-on label).

A0029214

16

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

5.2.2

Storage and transport

Measuring devices with lifting lugs

LCAUTION
Special transportation instructions for devices with lifting lugs
‣ Only use the lifting lugs fitted on the device or flanges to transport the device.
‣ The device must always be secured at two lifting lugs at least.

5.2.3

Transporting with a fork lift

If transporting in wood crates, the floor structure enables the crates to be lifted lengthwise
or at both sides using a forklift.

5.3

Packaging disposal

All packaging materials are environmentally friendly and 100 % recyclable:• Outer packaging of device
Polymer stretch wrap, complying with EU Directive 2002/95/EC (RoHS)• Packaging• Wooden crate treated in accordance with ISPM 15 standard, confirmed by IPPC logo
• Cardboard box in accordance with European packaging guideline 94/62EC,
recyclability confirmed by Resy symbol• Carrying and securing materials• Disposable plastic pallet• Plastic straps• Plastic adhesive strips• Filler material
Paper pads

Endress+Hauser

17

Mounting

Proline Promass P 100 PROFIBUS DP

6

Mounting

6.1

Mounting requirements

6.1.1

Mounting position

Mounting location

A0028772

To prevent measuring errors arising from accumulation of gas bubbles in the measuring
pipe, avoid the following mounting locations in the piping:• Highest point of a pipeline.
• Directly upstream of a free pipe outlet in a down pipe.
Installation in down pipes
However, the following installation suggestion allows for installation in an open verticalpipeline. Pipe restrictions or the use of an orifice with a smaller cross-section than thenominal diameter prevent the sensor running empty while measurement is in progress.

1

2

3
4

5
A0028773

4
1
2
3
4
5

18

Installation in a down pipe (e.g. for batching applications)
Supply tankSensorOrifice plate, pipe restriction
Valve
Batching tankEndress+Hauser

Proline Promass P 100 PROFIBUS DP

MountingDNØ orifice plate, pipe restriction

[mm]

[in]

[mm]

[in]

8

³⁄₈

6

0.24

15

½

10

0.40

25

1

14

0.55

40

1½

22

0.87

50

2

28

1.10

Orientation
The direction of the arrow on the sensor nameplate helps you to install the sensor
according to the flow direction (direction of medium flow through the piping).
Orientation
A

Recommendation
1)

Vertical orientation

A0015591

B

2)

Horizontal orientation, transmitter at
top

Exception:→  5,  19
A0015589

C

3)

Horizontal orientation, transmitter atbottomException:→  5,  19
A0015590

D

Horizontal orientation, transmitter atside
A0015592

1)
2)3)This orientation is recommended to ensure self-draining.
Applications with low process temperatures may reduce the ambient temperature. To maintain theminimum ambient temperature for the transmitter, this orientation is recommended.
Applications with high process temperatures may increase the ambient temperature. To maintain the
maximum ambient temperature for the transmitter, this orientation is recommended.

If a sensor is installed horizontally with a curved measuring tube, match the position of thesensor to the fluid properties.

1

2
A0028774

5
1
2

Endress+Hauser

Orientation of sensor with curved measuring tube
Avoid this orientation for fluids with entrained solids: Risk of solids accumulating.
Avoid this orientation for outgassing fluids: Risk of gas accumulating.

19

Mounting

Proline Promass P 100 PROFIBUS DP

Inlet and outlet runs
No special precautions need to be taken for fittings that create turbulence, such as valves,elbows or T-pieces, as long as no cavitation occurs →  20.

A0029322

A0029323

Dimensions
For the dimensions and installation lengths of the device, see the "Technical
Information" document, "Mechanical construction" section .

6.1.2

Environment and process requirements

Ambient temperature range
Measuring device• –40 to +60 °C (–40 to +140 °F)• Order code for "Test, certificate", option JM:–50 to +60 °C (–58 to +140 °F)‣ If operating outdoors:
Avoid direct sunlight, particularly in warm climatic regions.
System pressure
It is important that cavitation does not occur, or that gases entrained in the liquids do not
outgas.
Cavitation is caused if the pressure drops below the vapor pressure:• In liquids that have a low boiling point (e.g. hydrocarbons, solvents, liquefied gases)• In suction lines‣ Ensure the system pressure is sufficiently high to prevent cavitation and outgassing.
For this reason, the following mounting locations are recommended:• At the lowest point in a vertical pipe
• Downstream from pumps (no danger of vacuum)

A0028777

Thermal insulation
In the case of some fluids, it is important to keep the heat radiated from the sensor to thetransmitter to a low level. A wide range of materials can be used for the required
insulation.

20

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Mounting

The following device versions are recommended for versions with thermal insulation:• Version with extended neck for insulation:
Order code for "Sensor option", option CG with an extended neck length of
105 mm (4.13 in).
• Extended temperature version:
Order code for "Measuring tube material", option TD or TG with an extended neck lengthof 105 mm (4.13 in).
NOTICE
Electronics overheating on account of thermal insulation!
‣ Recommended orientation: horizontal orientation, transmitter housing pointing
downwards.
‣ Do not insulate the transmitter housing .
‣ Maximum permissible temperature at the lower end of the transmitter housing:
80 °C (176 °F)‣ Thermal insulation with not isolated extended neck: We recommend that you do not
insulate the extended neck in order to ensure optimum dissipation of heat.

A0034391

6

Thermal insulation with not isolated extended neckHeating
NOTICE
Electronics can overheat due to elevated ambient temperature!
‣ Observe maximum permitted ambient temperature for the transmitter.
‣ Depending on the medium temperature, take the device orientation requirements into
account.
NOTICE
Danger of overheating when heating‣ Ensure that the temperature at the lower end of the transmitter housing does notexceed 80 °C (176 °F).
‣ Ensure that sufficient convection takes place at the transmitter neck.
‣ Ensure that a sufficiently large area of the transmitter neck remains exposed. The
uncovered part serves as a radiator and protects the electronics from overheating andexcessive cooling.
‣ When using in potentially explosive atmospheres, observe the information in thedevice-specific Ex documentation. For detailed information on the temperature tables,
see the separate document entitled "Safety Instructions" (XA) for the device.
‣ Take account of the behavior of process diagnostic "830 Ambient temperature too high"and "832 Electronics temperature too high" if it is not possible to avoid overheating witha suitable system layout.

Endress+Hauser

21

Mounting

Proline Promass P 100 PROFIBUS DP

Heating options
If a fluid requires that no heat loss should occur at the sensor, users can avail of thefollowing heating options:• Electrical heating, e.g. with electric band heaters 1)• Via pipes carrying hot water or steam• Via heating jackets
Vibrations
The high oscillation frequency of the measuring tubes ensures that the correct operation of
the measuring system is not influenced by plant vibrations.

6.1.3

Special mounting instructions

Drainability
When installed vertically, the measuring tube can be drained completely and protectedagainst buildup.
When the sensor is installed in a horizontal line, eccentric clamps can be used to ensurecomplete drainability. When the system is pitched in a specific direction and at a specificslope, gravity can be used to achieve complete drainability. The sensor must be mounted in
the correct position to ensure full drainability in the horizontal position. Markings on thesensor show the correct mounting position to optimize drainability.

2
1
1

5

4
3
A0016583

1
2
3
4
5

Eccentric clamp connection"This side up" label indicates which side is up
For DN 8 to 25(3/8 to 1"): Gradient: approx. 2% or 21 mm/m (0.24 in/ft); for DN 40 to 50(1½ to 2"):
Gradient approx. 2° or 35 mm/m (0.42 in/ft)
TransmitterLine on the underside indicates the lowest point of the eccentric process connection.

Hygienic compatibility
When installing in hygienic applications, please refer to the information in the"Certificates and approvals/hygienic compatibility" section →  145
Securing with mounting clamp in the case of hygiene connections
It is not necessary to provide additional support for the sensor for operational performance
purposes. If, however, additional support is required for installation purposes, thefollowing dimensions must be observed.
1)

22

The use of parallel electric band heaters is generally recommended (bidirectional electricity flow). Particular considerations must be made if asingle-wire heating cable is to be used. Additional information is provided in the document EA01339D "Installation instructions for electrical
trace heating systems".

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Mounting

Use mounting clamp with lining between clamp and measuring instrument.

C

B

A
A0030298

DN

A

B

C

[mm]

[in]

[mm]

[in]

[mm]

[in]

[mm]

[in]

8

³⁄₈

298

11.73

33

1.3

28

1.1

15

½

402

15.83

33

1.3

28

1.1

25

1

542

21.34

33

1.3

38

1.5

40

1½

658

25.91

36.5

1.44

56

2.2

50

2

772

30.39

44.1

1.74

75

2.95

6.2

Mounting the measuring device

6.2.1

Required tool

For sensor
For flanges and other process connections: use a suitable mounting tool

6.2.2

Preparing the measuring device

1.

Remove all remaining transport packaging.

2.

Remove any protective covers or protective caps present from the sensor.

3.

Remove stick-on label on the electronics compartment cover.

6.2.3

Mounting the measuring device

LWARNING
Danger due to improper process sealing!
‣ Ensure that the inside diameters of the gaskets are greater than or equal to that of theprocess connections and piping.
‣ Ensure that the seals are clean and undamaged.
‣ Secure the seals correctly.
1.

Endress+Hauser

Ensure that the direction of the arrow on the nameplate of the sensor matches theflow direction of the medium.

23

Mounting

Proline Promass P 100 PROFIBUS DP

2.

Install the measuring device or turn the transmitter housing so that the cable entries
do not point upwards.
A0029263

6.2.4

Turning the display module

The local display is only available with the following device version:
Order code for "Display; Operation", option B: 4-line; lit, via communication
The display module can be turned to optimize display readability.
Aluminum housing version, AlSi10Mg, coated
3.
5.

2.
7.
4.

6.

4.

4.
1.
3 mm
A0023192

Compact and ultra-compact housing version, hygienic, stainless

1.

8 mm

5.
4.

2.
3.

3.
6.

3.

3.

A0023195

24

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

6.3

Mounting

Post-installation check

Is the device undamaged (visual inspection)?



Does the measuring device conform to the measuring point specifications?
For example:• Process temperature →  137• Process pressure (refer to the section on "Pressure-temperature ratings" in the "Technical
Information" document)• Ambient temperature• Measuring range

Has the correct orientation for the sensor been selected ?
• According to sensor type
• According to medium temperature• According to medium properties (outgassing, with entrained solids)

Endress+Hauser



Does the arrow on the sensor nameplate match the direction of flow of the fluid through thepiping →  19?

Are the measuring point identification and labeling correct (visual inspection)?

Is the device adequately protected from precipitation and direct sunlight?



Are the securing screw and securing clamp tightened securely?



25

Electrical connection

Proline Promass P 100 PROFIBUS DP

7

Electrical connection

LWARNING
Live parts! Incorrect work performed on the electrical connections can result in anelectric shock.
‣ Set up a disconnecting device (switch or power-circuit breaker) to easily disconnect thedevice from the supply voltage.
‣ In addition to the device fuse, include an overcurrent protection unit with max. 16 A in
the plant installation.

7.1

Electrical safety

In accordance with applicable national regulations.

7.2

Connecting requirements

7.2.1

Required tools• For cable entries: Use corresponding tools• For securing clamp (on aluminum housing): Allen screw3 mm• For securing screw (for stainless steel housing): open-ended wrench 8 mm
• Wire stripper• When using stranded cables: crimper for wire end ferrule

7.2.2

Requirements for connecting cable

The connecting cables provided by the customer must fulfill the following requirements.
Permitted temperature range• The installation guidelines that apply in the country of installation must be observed.
• The cables must be suitable for the minimum and maximum temperatures to be
expected.
Power supply cable (incl. conductor for the inner ground terminal)
Standard installation cable is sufficient.
Signal cable
PROFIBUS DP
The IEC 61158 standard specifies two types of cable (A and B) for the bus line which canbe used for every transmission rate. Cable type A is recommended.

26

Cable typeA

Characteristic impedance

135 to 165 Ω at a measuring frequency of 3 to 20 MHz

Cable capacitance< 30 pF/mWire cross-section> 0.34 mm2 (22 AWG)

Cable type

Twisted pairs

Loop resistance≤110 Ω/km

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Electrical connection

Signal damping

Max. 9 dB over the entire length of the cable cross-section

ShieldCopper braided shielding or braided shielding with foil shield. When grounding
the cable shield, observe the grounding concept of the plant.

For further information on planning and installing PROFIBUS networks see:Operating Instructions "PROFIBUS DP/PA: Guidelines for planning and commissioning"(BA00034S)
Cable diameter• Cable glands supplied:
M20 × 1.5 with cable ⌀ 6 to 12 mm (0.24 to 0.47 in)• Spring terminals:
Wire cross-sections 0.5 to 2.5 mm2 (20 to 14 AWG)

7.2.3

Terminal assignment

Transmitter
PROFIBUS DP connection version
For use in the non-hazardous area and Zone 2/Div. 2
Order code for "Output", option L
Depending on the housing version, the transmitters can be ordered with terminals or
device plugs.
Order code
"Housing"

Connection methods available

Possible options for order code
"Electrical connection"

OutputPowersupplyOptionsA, B

Terminals

Terminals• Option A: coupling M20x1• Option B: thread M20x1• Option C: thread G ½"
• Option D: thread NPT ½"

Options
A, B

Device plugconnectors→  28Terminals• Option L: plug M12x1 + thread NPT ½"• Option N: plug M12x1 + coupling M20• Option P: plug M12x1 + thread G ½"
• Option U: plug M12x1 + thread M20

Options
A, B, C

Device plugconnectors→  28Device plugconnectors→  28

Option Q: 2 x plug M12x1

Order code for "Housing":• Option A: compact, coated aluminum• Option B: compact, hygienic, stainless• Option C ultra-compact, hygienic, stainlessEndress+Hauser

27

Electrical connection

Proline Promass P 100 PROFIBUS DP

26 B
27 A

2

1 L+
2 L_

1

A0022716

7
1
2

PROFIBUS DP terminal assignment
Power supply: DC 24 V
PROFIBUS DP

Terminal number
Order code
"Output"

Power supply
2 (L-)

Option L

Output1 (L+)

26 (RxD/TxD-P)

27 (RxD/TxDN)

BA

DC 24 V

Order code for "Output":
Option L: PROFIBUS DP, for use in non-hazardous areas and Zone 2/Div. 2

7.2.4

Pin assignment, device plug

Supply voltage
For use in the non-hazardous area and Zone 2/Div. 2.
Pin2

Assignment

1

1

3

5
4
A0016809

L+

DC 24 V2

Not assigned

3

Not assigned

4

L-

5

DC 24 V
Grounding/shielding

CodingPlug/socket

A

Plug

Device plug for signal transmission (device side)
Pin

2

Assignment

1

3

1

Not assigned

2

A

3

5
A0016811

28

Not assigned

4

4

PROFIBUS DP

B

5

PROFIBUS DP
Grounding/shielding

CodingPlug/socket

B

SocketEndress+Hauser

Proline Promass P 100 PROFIBUS DP

7.2.5

Electrical connection

Preparing the measuring device

NOTICE
Insufficient sealing of the housing!
Operational reliability of the measuring device could be compromised.
‣ Use suitable cable glands corresponding to the degree of protection.
1.

Remove dummy plug if present.

2.

If the measuring device is supplied without cable glands:
Provide suitable cable gland for corresponding connecting cable.

3.

If the measuring device is supplied with cable glands:
Observe requirements for connecting cables →  26.

7.3

Connecting the measuring device

NOTICEAn incorrect connection compromises electrical safety!
‣ Have electrical connection work carried out by appropriately trained specialists only.
‣ Observe applicable federal/national installation codes and regulations.
‣ Comply with local workplace safety regulations.
‣ Always connect the protective ground cable  before connecting additional cables.
‣ When using in potentially explosive atmospheres, observe the information in thedevice-specific Ex documentation.

7.3.1

Connecting the transmitter

The connection of the transmitter depends on the following order codes:• Housing version: compact or ultra-compact• Connection version: device plug or terminals

A

B
1

2

C
1

2

3

4
A0016924

Endress+Hauser

8

Housing versions and connection versions

A
B
1
2
C
3
4

Housing version: compact, coated, aluminum
Housing version: compact, hygienic, stainlessCable entry or device plug for signal transmission
Cable entry or device plug for supply voltage
Housing version: ultra-compact, hygienic, stainlessDevice plug for signal transmission
Device plug for supply voltage

29

Electrical connection

Proline Promass P 100 PROFIBUS DP

8 mm

8 mm

3 mm

2

2

2

1

1
1

2

3

3

10 (0.4)mm (in)A00178449123Device versions with connection examples
Cable
Device plug for signal transmission
Device plug for supply voltage

For device version with device plug: follow step 6 only.
1.

Depending on the housing version, loosen the securing clamp or fixing screw of the
housing cover.

2.

Depending on the housing version, unscrew or open the housing cover anddisconnect the local display from the main electronics module where necessary→  143.

3.

Push the cable through the cable entry . To ensure tight sealing, do not remove thesealing ring from the cable entry.

4.

Strip the cable and cable ends. In the case of stranded cables, also fit ferrules.

5.

Connect the cable in accordance with the terminal assignment or the device plug pin
assignment .

6.

Depending on the device version, tighten the cable glands or plug in the device plugand tighten .

7.

LWARNING
Housing degree of protection may be voided due to insufficient sealing of the
housing.
‣ Screw in the screw without using any lubricant. The threads on the cover arecoated with a dry lubricant.
Reverse the removal procedure to reassemble the transmitter.

30

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

7.3.2

Electrical connection

Potential equalization

Requirements
No special measures for potential equalization are required.
For devices intended for use in hazardous locations, please observe the guidelines inthe Ex documentation (XA).
Connection example, standard scenario
Connection example in special situations

7.4

Special connection instructions

7.4.1

Connection examples

PROFIBUS DP
1

2

3

A
B

A
4
B

A
4
B

A0028765

 10
1
2
3
4

Connection example for PROFIBUS DP, non-hazardous area and Zone 2/Div. 2

Control system (e.g. PLC)
Cable shield provided at one end. The cable shield must be grounded at both ends to comply with EMC
requirements; observe cable specifications
Distribution box
Transmitter

If baud rates > 1.5 MBaud an EMC cable entry must be used and the cable shield must
continue as far as the terminal wherever possible.

7.5

Hardware settings

7.5.1

Setting the device address

PROFIBUS DP
The address must always be configured for a PROFIBUS DP/PA device. The valid address
range is between 1 and 126. In a PROFIBUS DP/PA network, each address can only be
assigned once. If an address is not configured correctly, the device is not recognized by themaster. All measuring devices are delivered from the factory with the device address 126
and with the software addressing method.

Endress+Hauser

31

Electrical connection

Proline Promass P 100 PROFIBUS DP

Setting the address

1

2

2

3

4

4

8

5

16

6

32

7

64

PROFIBUS
address

OFF ON

1

8- Software addressing

9- Write protection

10- Not used

A0021265

 11

Addressing using DIP switches on the I/O electronics module1.

Depending on the housing version, loosen the securing clamp or fixing screw of the
housing cover.

2.

Depending on the housing version, unscrew or open the housing cover anddisconnect the local display from the main electronics module where necessary→  143.

3.

Disable software addressing via DIP switch 8 (OFF).

4.

Set the desired device address via the corresponding DIP switches.
 Example →  11,  32: 1 + 16 + 32 = device address 49
The device demands rebooting after 10 s. After rebooting, hardware addressing
is enabled with the configured IP address.

5.

Reverse the removal procedure to reassemble the transmitter.

7.5.2

Enabling the terminating resistor

PROFIBUS DP
To avoid incorrect communication transmission caused by impedance mismatch, terminate
the PROFIBUS DP cable correctly at the start and end of the bus segment.
• If the device is operated with a baud rate of 1.5 MBaud and under:
For the last transmitter on the bus, terminate via DIP switch 2 (bus termination) and DIP
switch 1 and 3 (bus polarization). Setting: ON – ON – ON →  12,  33.
• For baud rates > 1.5 MBaud:Due to the capacitance load of the user and the line reflections generated as a result,
ensure that an external bus terminator is used.
It is generally advisable to use an external bus terminator as the entire segment canfail if a device that is terminated internally is defective.

32

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Electrical connection

3.
2.

1.

5V

4.

390 Ω
DIP 1

OFF ON1

Bus polarisation2

Bus termination

3

Bus polarisation

4

Not used

26

DIP 2
220 Ω
27

DIP 3
390 Ω
0V
A0021274

 12

7.6

Termination using DIP switches on the I/O electronics module (for baud rates < 1.5 MBaud)

Ensuring the degree of protection

The measuring device fulfills all the requirements for the degree of protection IP66/67,
Type 4X enclosure.
To guarantee the degree of protection IP66/67, Type 4X enclosure, carry out the following
steps after the electrical connection:
1.

Check that the housing seals are clean and fitted correctly.

2.

Dry, clean or replace the seals if necessary.

3.

Tighten all housing screws and screw covers.

4.

Firmly tighten the cable glands.

5.

To ensure that moisture does not enter the cable entry:
Route the cable so that it loops down before the cable entry ("water trap").
A00292786.

7.7

Endress+Hauser

The cable glands supplied do not provide any housing protection if they are not used.
Therefore, they must be replaced by dummy plugs that match the housing protection.

Post-connection check

Are cables or the device undamaged (visual inspection)?



Do the cables used comply with the requirements →  26?

33

Electrical connection

34

Proline Promass P 100 PROFIBUS DP

Do the mounted cables have adequate strain relief?



Are all cable glands installed, securely tightened and leak-tight? Cable run with "water trap"→  33?



Depending on the device version: are all the device plugs firmly tightened →  29?

Does the supply voltage match the specifications on the transmitter nameplate →  132?

Is the terminal assignment →  27 or the device plug pin assignment →  28 correct?



If supply voltage is present, is the power LED on the electronics module of the transmitter lit green→  11?



Depending on the device version, is the securing clamp or fixing screw firmly tightened?

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Operation options

8

Operation options

8.1

Overview of operating options

1

2
A0017760

12

Endress+Hauser

Computer with Web browser (e.g. Internet Explorer) or with "FieldCare" operating tool
Automation system, e.g. "RSLogix" (Rockwell Automation) and work station for measuring device operation
with Add-on Profile Level 3 for "RSLogix 5000" software (Rockwell Automation)

35

Operation options

Proline Promass P 100 PROFIBUS DP

8.2

Structure and function of the operating menu

8.2.1

Structure of the operating menu

For an overview of the operating menu for experts: see the "Description of Device
Parameters" document supplied with the device →  148

Operating menu for operators and maintenances
Language
Operatation!
Language

Operator

Parameter 1
Parameter n
Submenu 1
Submenu n
Setup

Device tag

Wizard n / Parameter nAdvanced setup

Enter access code
Parameter 1

Task-oriented

Maintenance

Wizard 1 / Parameter 1

Parameter n
Submenu 1
Submenu n
Diagnostics

Parameter 1
Parameter n
Submenu 1
Submenu nOperating menu for experts
ExpertAccess status display
Parameter n

Sensor
Input
OutputCommunication

Function-oriented

ExpertSystemApplication
Diagnostics

A0018237-EN

 13

36

Schematic structure of the operating menu

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

8.2.2

Operation options

Operating philosophy

The individual parts of the operating menu are assigned to certain user roles (operator,
maintenance etc.). Each user role contains typical tasks within the device lifecycle.
Menu/parameter
Language

Taskoriented

Operation
Setup

User role and tasks

Content/meaning

Role "Operator", "Maintenance"
Tasks during operation:• Configuration of the operational
display• Reading measured values• Defining the operating language• Defining the Web server operating language• Resetting and controlling totalizers"Maintenance" roleCommissioning:
Configuration of the measurement

Submenus for fast commissioning:• Configuration of the system units• Definition of the medium• Configuration of the operational display• Configuration of the low flow cut off• Configuration of the detection of partially filled and empty pipes• Configuring the operational display (e.g. display format, display contrast)• Resetting and controlling totalizers

Advanced setup• For more customized configuration of the measurement (adaptation to
special measuring conditions)• Configuration of totalizers• Administration (define access code, reset measuring device)
Diagnostics

ExpertFunctionoriented

Endress+Hauser"Maintenance" role
Troubleshooting:• Diagnostics and elimination of process
and device errors• Measured value simulation

Contains all parameters for error detection and analyzing process and device
errors:• Diagnostic list
Contains up to 5 currently pending diagnostic messages.
• Event logbook
Contains event messages that have occurred.
• Device information
Contains information for identifying the device.
• Measured values
Contains all current measured values.
• Analog inputs
Is used to display the analog input.
• Heartbeat
The functionality of the device is checked on demand and the verification
results are documented.
• Simulation
Is used to simulate measured values or output values.

Tasks that require detailed knowledge ofthe function of the device:• Commissioning measurements underdifficult conditions• Optimal adaptation of themeasurement to difficult conditions• Detailed configuration of thecommunication interface• Error diagnostics in difficult cases

Contains all the parameters of the device and makes it possible to accessthese parameters directly using an access code. The structure of this menu isbased on the function blocks of the device:• SystemContains all higher-level device parameters that do not pertain either tothe measurement or to measured value communication.
• Sensor
Configuration of the measurement.
• Communication
Configuration of the digital communication interface and the Web server.
• Submenus for function blocks (e.g. "Analog Inputs")
Configuration of function blocks.
• Application
Configuration of the functions that go beyond the actual measurement(e.g. totalizer).
• Diagnostics
Error detection and analysis of process and device errors and for device
simulation and Heartbeat Technology.

37

Operation options

Proline Promass P 100 PROFIBUS DP

8.3

Displaying the measured values via the local display(optionally available)

8.3.1

Operational display

The local display is optionally available:
Order code for "Display; operation", option B ""4-line, illuminated; via communication".
1
2

F

XXXXXXXXX

4

31120.50kg/h
A0037831

12
34Operational display
Tag name
Status area
Display area for measured values (4-line)

Status area
The following symbols appear in the status area of the operational display at the top right:• Status signals• F: Failure• C: Function check• S: Out of specification• M: Maintenance required• Diagnostic behavior• : Alarm• : Warning• : Locking (the device is locked via the hardware )• : Communication (communication via remote operation is active)
Display area
In the display area, each measured value is prefaced by certain symbol types for further
description:
Measured variable

Measurement channel
numberDiagnostic behavior

↓

↓

↓

ExampleAppears only if a diagnostics
event is present for thismeasured variable.

Measured variables
SymbolMeaning
Mass flow• Volume flow
• Corrected volume flow

38

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Operation options• Density• Reference density
Temperature
Totalizer
The measurement channel number indicates which of the three totalizers is displayed.

Measurement channel numbers
SymbolMeaning
Measurement channel 1 to 4

The measurement channel number is displayed only if more than one channel is present for the same measured
variable type (e.g. Totalizer 1 to 3).

Diagnostic behavior
The diagnostic behavior pertains to a diagnostic event that is relevant to the displayed measured variable.
For information on the symbols

The number and display format of the measured values can only be configured via thecontrol system or Web server.

8.3.2

User roles and related access authorization

The two user roles "Operator" and "Maintenance" have different write access to theparameters if the customer defines a user-specific access code. This protects the device
configuration from unauthorized access .
Defining access authorization for user roles
An access code is not yet defined when the device is delivered from the factory. Accessauthorization (read and write access) to the device is not restricted and corresponds to the"Maintenance" user role.

‣ Define the access code.
 The "Operator" user role is redefined in addition to the "Maintenance" user role.
Access authorization differs for the two user roles.
Access authorization to parameters: "Maintenance" user role
Access code status

Read access

Write accessAn access code has not yet been defined(factory setting).



After an access code has been defined.



 1)

1)

The user only has write access after entering the access code.

Access authorization to parameters: "Operator" user role
Access code status
After an access code has been defined.
1)

Read access

Write access



– 1)

Despite the defined access code, certain parameters can always be modified and thus are excluded fromthe write protection, as they do not affect the measurement. Refer to the "Write protection via access code"
sectionThe user role with which the user is currently logged on is indicated by the .
Navigation path:
Endress+Hauser

39

Operation options

Proline Promass P 100 PROFIBUS DP

8.4

Access to operating menu via web browser

8.4.1

PROFINET with Ethernet-APL

Device use

Device connection to an APL field switch
The device may only be operated according to the following APL port
classifications:• If used in hazardous areas: SLAA or SLAC 1)• If used in non-hazardous areas: SLAX
• Connection values of APL field switch (for example corresponds to APL port
classification SPCC or SPAA):• Maximum input voltage: 15 VDC• Minimum output values: 0.54 W
Device connection to an SPE switch
If used in non-hazardous areas: suitable SPE switch
SPE switch prerequisite:• Support of 10BASE-T1L standard• Support of PoDL power class 10, 11 or 12• Detection of SPE field devices without integrated PoDL module
Connection values of SPE switch:• Maximum input voltage: 30 VDC• Minimum output values: 1.85 W

PROFINET

According to IEC 61158 and IEC 61784

Ethernet-APL

According to IEEE 802.3cg, APL port profile specification v1.0, galvanicallyisolated

Data transfer

10 Mbit/s

Current consumption

TransmitterMax. 55.56 mA

Permitted supply voltage• Ex: 9 to 15 V
• Non-Ex: 9 to 32 V

Network connection

With integrated reverse polarity protection

1)

For more information on using the device in the hazardous area, see the Ex-specific Safety Instructions

8.4.2

Prerequisites

Computer hardware
Hardware40

Interface
CDI-RJ45

WLAN

Interface

The computer must have an RJ45
interface.

The operating unit must have a
WLAN interface.

Connection

Standard Ethernet cable with RJ45
connector.

Connection via Wireless LAN.

Display

Recommended size: ≥12" (depends on the screen resolution)

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Operation options

Computer software
Software

Interface
CDI-RJ45

Recommended operating
systems

WLAN• Microsoft Windows 8 or higher.
• Mobile operating systems:• iOS• Android Microsoft Windows XP is supported.
 Microsoft Windows 7 is supported.
Web browsers supported• Microsoft Internet Explorer 8 or higher• Microsoft Edge
• Mozilla Firefox• Google Chrome• SafariComputer settings
User rights

Appropriate user rights (e.g. administrator rights) for TCP/IP and proxy server
settings are necessary (for adjusting the IP address, subnet mask etc.).

Proxy server settings of the
Web browser

The web browser setting Use a Proxy Server for Your LAN must be deselected .

JavaScript

JavaScript must be enabled.
JavaScript cannot be enabled: IfEnter
http://XXX.XXX.X.X.XX/servlet/basic.html in the address bar of
the web browser, e.g. http://192.168.1.212/servlet/basic.html. A fully
functional but simplified version of the operating menu structure starts
in the web browser.

Network connections

Only the active network connections to the measuring device should be used.
Switch off all other network connections.

In the event of connection problems: →  83
Measuring device: Via CDI-RJ45 service interface
Device

CDI-RJ45 service interface

Measuring device

The measuring device has an RJ45 interface.

Web server

Web server must be enabled; factory setting: ON

 For information on enabling the Web server →  44
8.4.3

Establishing a connection

Via service interface (CDI-RJ45)
Preparing the measuring device
Configuring the Internet protocol of the computer
The following information refers to the default Ethernet settings of the device.
IP address of the device: 192.168.1.212 (factory setting)Endress+Hauser

1.

Switch on the measuring device.

2.

Connect the computer to the RJ45 plug via the standard Ethernet cable →  144.

41

Operation options

Proline Promass P 100 PROFIBUS DP

3.

If a 2nd network card is not used, close all the applications on the notebook.
 Applications requiring Internet or a network, such as e-mail, SAP applications,Internet or Windows Explorer.

4.

Close any open Internet browsers.

5.

Configure the properties of the Internet protocol (TCP/IP) as defined in the table:

IP address

192.168.1.XXX; for XXX all numerical sequences except: 0, 212 and 255 → e.g.
192.168.1.213

Subnet mask

255.255.255.0

Default gateway

192.168.1.212 or leave cells empty

Starting the Web browser
1.

Start the Web browser on the computer.

2.

Enter the IP address of the web server in the address line of the web browser:
192.168.1.212
 The login page appears.
1

2

3

4

5

6

7
8
9
10
A0029417

1
2
3
4
5
6
7
8
9
10

Picture of device
Device name
Device tag
Status signal
Current measured values
Operating language
User role
Access code
Login
Reset access code

If a login page does not appear, or if the page is incomplete →  83

8.4.4

42

Logging on

1.

Select the preferred operating language for the Web browser.

2.

Enter the user-specific access code.

3.

Press OK to confirm your entry.

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Operation options

Access code

0000 (factory setting); can be changed by customer

If no action is performed for 10 minutes, the Web browser automatically returns to
the login page.

8.4.5

User interface
1

2

6

3

4

5

7
A0032879

1
2
3
4
5
6
7

Picture of device
Device nameTag name
Status signal
Current measured values
Navigation area
Local display language

Header
The following information appears in the header:• Device name• Device tag• Device status with status signal →  85
• Current measured values
Function row
Functions

Meaning

Measured values Displays the measured values of the device

Menu

Device status

Endress+Hauser• Access to the operating menu from the measuring device• The structure of the operating menu is the same as for the operating tools
Detailed information on the structure of the operating menu: see the Description of Device
Parameters
Displays the diagnostic messages currently pending, listed in order of priority

43

Operation options

Proline Promass P 100 PROFIBUS DP

Functions

MeaningDatamanagement

Data exchange between computer and measuring device:• Device configuration:• Load settings from the device(XML format, save configuration)• Save settings to the device(XML format, restore configuration)• Logbook - Export Event logbook (.csv file)• Documents - Export documents:• Export backup data record(.csv file, create documentation of the measuring point configuration)• Verification report(PDF file, only available with the "Heartbeat Verification" application package)• File for system integration - If using fieldbuses, upload device drivers for system
integration from the measuring device:
PROFIBUS DP: GSD file

Network

Configuration and checking of all the parameters required for establishing the connection to
the measuring device:• Network settings (e.g. IP address, MAC address)• Device information (e.g. serial number, firmware version)

Logout

End the operation and call up the login page

Navigation area
The menus, the associated submenus and parameters can be selected in the navigationarea.
Working area
Depending on the selected function and the related submenus, various actions can be
performed in this area:• Configuring parameters• Reading measured values• Calling up help text
• Starting an upload/download

8.4.6

Disabling the Web server

The Web server of the measuring device can be switched on and off as required using the
Web server functionality parameter.
Navigation"Expert" menu → Communication → Web server
Parameter overview with brief description
Parameter
Web server functionality

Description
Switch the Web server on and off.

Selection
• Off
• On

Function scope of the "Web server functionality" parameter

44

OptionDescription

Off• The Web server is completely disabled.
• Port 80 is locked.

On• The complete Web server functionality is available.
• JavaScript is used.
• The password is transferred in an encrypted state.
• Any change to the password is also transferred in an encrypted state.

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Operation options

Enabling the Web server
If the Web server is disabled it can only be re-enabled with the Web server functionality
parameter via the following operating options:• Via Bedientool "FieldCare"• Via "DeviceCare" operating tool

8.4.7

Logging out

Before logging out, perform a data backup via the Data management function(upload configuration from device) if necessary.
1.

Select the Logout entry in the function row.
 The home page with the Login box appears.

2.

Close the Web browser.

3.

If no longer needed:
Reset the modified properties of the Internet protocol (TCP/IP) →  41.

8.5

Access to the operating menu via the operating tool

8.5.1

Connecting the operating tool

Via PROFIBUS DP network
This communication interface is available in device versions with PROFIBUS DP.
1

2

3

4

4

4
A0020903

 14
1
2
3
4

Options for remote operation via PROFIBUS DP network

Automation system
Computer with PROFIBUS network card
PROFIBUS DP network
Measuring device

Via service interface (CDI-RJ45)

Endress+Hauser

45

Operation options

Proline Promass P 100 PROFIBUS DPPROFIBUS DP23

1
A0021270

 15
1
2
3

Connection for order code for "Output", option L: PROFIBUS DP

Service interface (CDI-RJ45) of the measuring device with access to the integrated web server
Computer with web browser (e.g. Internet Explorer) for accessing the integrated web server or with "FieldCare"
operating tool with COM DTM "CDI Communication TCP/IP"
Standard Ethernet connecting cable with RJ45 plug

8.5.2

FieldCare

Function scope
FDT (Field Device Technology)-based plant asset management tool from Endress+Hauser.
It can configure all smart field units in a system and helps you manage them. By using thestatus information, it is also a simple but effective way of checking their status andcondition.
Access is via:
CDI-RJ45 service interface
Typical functions:• Configuring parameters of transmitters• Loading and saving device data (upload/download)• Documentation of the measuring point• Visualization of the measured value memory (line recorder) and event logbook
For additional information about FieldCare, see Operating Instructions BA00027S
and BA00059S
Source for device description files
See information →  48
Establishing a connection46

1.

Start FieldCare and launch the project.

2.

In the network: Add a device.
 The Add device window opens.

3.

Select the CDI Communication TCP/IP option from the list and press OK to confirm.

4.

Right-click CDI Communication TCP/IP and select the Add device option in thecontext menu that opens.

5.

Select the desired device from the list and press OK to confirm.
 The CDI Communication TCP/IP (Configuration) window opens.

6.

Enter the device address in the IP address field and press Enter to confirm:
192.168.1.212 (factory setting); if the IP address is not known .
Endress+Hauser

Proline Promass P 100 PROFIBUS DP

7.

Operation options

Establish the online connection to the device.
For additional information, see Operating Instructions BA00027S and BA00059S

User interface
2

4

3

6

5

7

Xxxxxx/…/…/

1

Device name:

Xxxxxxx

Mass flow:

12.34

kg/h

Device tag:

Xxxxxxx

Volume flow:

12.34

m³/h

Status:Good

XxxxxxAccess status tooling+
Operation
P

–

Setup
Device tag–
System unitsP
Mass flow unit
P
Volume flow unit+
Select medium
P

8

+

…

+

…

+

Maintenance

Mass flow unit:

kg/hVolume flow unit:m³/h

Xxxxxx
kg/h
m³/h

9

Advanced setup+

Diagnostics+

Expert10

11A0021051-EN

1
2
3
4
5
6
7
8
9
10
11

HeaderPicture of device
Device nameTag name
Status area with status signal →  85
Display area for current measured values
Edit toolbar with additional functions such as save/load, event list and create documentation
Navigation area with operating menu structure
Working area
Range of action
Status area

8.5.3

DeviceCare

Function scope
Tool to connect and configure Endress+Hauser field devices.
The fastest way to configure Endress+Hauser field devices is with the dedicated"DeviceCare" tool. Together with the device type managers (DTMs) it presents a convenient,comprehensive solution.
For details, see Innovation Brochure IN01047S
Source for device description files
See information →  48

Endress+Hauser

47

System integration

Proline Promass P 100 PROFIBUS DP

9System integration

9.1

Overview of device description files

9.1.1

Current version data for the device

Firmware version

01.01.zz• On the title page of the Operating Instructions• On the transmitter nameplate• Firmware version parameter parameter
Diagnostics → Device information → Firmware
versionRelease date of firmware version

10.2014---

Manufacturer ID

0x11

Manufacturer ID parameter parameter
Diagnostics → Device information → Manufacturer ID

Device type ID

0x1561

Device type parameter parameter
Diagnostics → Device information → Device type

Profile version

3.02

---

For an overview of the various firmware versions for the device

9.1.2

Operating tools

The suitable device description file for the individual operating tools is listed in the tablebelow, along with information on where the file can be acquired.
Operating tool via
PROFIBUS protocol

Sources for obtaining device descriptions

FieldCare• www.endress.com → Download Area
• USB stick (contact Endress+Hauser)• DVD (contact Endress+Hauser)

DeviceCare• www.endress.com → Download Area
• CD–ROM (contact Endress+Hauser)• DVD (contact Endress+Hauser)

9.2

Device master file (GSD)

In order to integrate field devices into a bus system, the PROFIBUS system needs a
description of the device parameters, such as output data, input data, data format, data
volume and supported transmission rate.
These data are available in the device master file (GSD) which is provided to the PROFIBUS
master when the communication system is commissioned. In addition device bit maps,
which appear as icons in the network structure, can also be integrated.
With the Profile 3.02 device master file (GSD) it is possible to exchange field devices madeby different manufacturers without having to reconfigure.
Generally speaking, it is possible to use two different GSDs with Profile 3.02 and higher:
the manufacturer-specific GSD and the Profile GSD.
• Before configuring, the user must decide which GSD should be used to operate thesystem.
• The setting can be changed via a class 2 master.

48

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

9.2.1

System integration

Manufacturer-specific GSD

This GSD guarantees the unrestricted functionality of the measuring device. Device-specificprocess parameters and functions are therefore available.
Manufacturer-specific GSD

ID numberFile name

PROFIBUS DP

0x1561

EH3x1561.gsd

The fact that the manufacturer-specific GSD should be used is specified in the Ident
number selector parameter by selecting the Manufacturer option.
Where to acquire the manufacturer-specific GSD:
www.endress.com → Download Area

9.2.2

Profile GSD

Differs in terms of the number of Analog Input blocks (AI) and the measured values. If asystem is configured with a Profile GSD, it is possible to exchange devices made bydifferent manufacturers. However, it is essential to ensure that the order of the cyclic
process values is correct.
ID number

Supported blocks

Supported channels

0x9740• 1 Analog Input• 1 Totalizer• Channel Analog Input: volume flow
• Channel totalizer: volume flow

0x9741• 2 Analog Input• 1 Totalizer• Channel Analog Input 1: volume flow
• Channel Analog Input 2: mass flow
• Channel totalizer: volume flow

0x9742• 3 Analog Input• 1 Totalizer• Channel Analog Input 1: volume flow
• Channel Analog Input 2: mass flow
• Channel Analog Input 3: corrected volume
flow• Channel totalizer: volume flow

The Profile GSD that is to be used is specified in the Ident number selector parameter byselecting the Profile 0x9740 option, Profile 0x9741 option or Profile 0x9742 option.

Endress+Hauser

49

System integration

Proline Promass P 100 PROFIBUS DP

9.3

Integration into a PROFIBUS network

9.3.1

Block model• Physical block• Function blocks• Analog Input Block• Analog Output Block• Discrete Input Block• Discrete Output Block• Totalizer Block
Technical values for the individual blocks →  131

9.3.2

Assignment of the measured values in the function blocks

The input value of a function block is defined via the CHANNEL parameter.
Analog Input 1 to 8 (AI)50

ChannelMeasured variable

33122

Volume flow

32961

Mass flow

33093

Corrected volume flow

708

Flow velocity

901

Target mass flow

793

Carrier mass flow

32850

Density

33092

Reference density

794

Concentration

1039

Dynamic viscosity

1032

Kinematic viscosity

904

Temperature compensated dynamic viscosity

905

Temperature compensated kinematic viscosity

33101

Temperature

263

Carrier pipe temperature

1042

Electronic temperature

1066

Oscillation frequency 01067

Oscillation frequency 1

1124

Oscillation amplitude 0876

Oscillation amplitude 11062

Frequency fluctuation 01063

Frequency fluctuation 1

1117

Oscillation damping 01118

Oscillation damping 11054

Tube damping fluctuation 01055

Tube damping fluctuation 1

1125

Signal asymmetry

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

System integration

ChannelMeasured variable

1056

Exciter current 0

1057

Exciter current 1

1440

Sensor integrity

Analog Output 1 to 3 (AO)
ChannelMeasured variable

306

External pressure 1)

307

External temperature

488

External reference density

1)

The compensation variables must be transmitted to the device in the SI basic unit.

The measured variable is accessed via Expert → Sensor → External compensation
Digital Input 1 to 2 (DI)ChannelSignal894

Empty pipe detection

895

Low flow cut off

1430

Verification status

Digital Output 1 to 3 (DO)
ChannelSignal890

Zero point adjustment891

Flow override

1429

Start verification

Totalizer 1 to 3 (TOT)

Endress+Hauser

ChannelSignal33122

Volume flow

32961

Mass flow

33093

Corrected volume flow

901

Target mass flow

793

Carrier mass flow

9.3.3

Totalizer control SET_TOT

Value

Behavior0

Totalize

1

Reset + hold2

Preset + hold51

System integration

Proline Promass P 100 PROFIBUS DP

9.4

Cyclic data transmission

Cyclic data transmission when using the device master file (GSD).

9.4.1

Block model

The block model shows which input and output data the measuring device makes available
for cyclic data exchange. Cyclic data exchange takes place with a PROFIBUS master (Class1), e.g. a control system.
Measuring device
Analog Input block 1 to 8

Totalizer block 1 to 3
TransducerBlock→  53→  54Control system
Output value AI→

Output value TOTAL→

Controller SETTOT←

Configuration MODETOT←

Analog Output block 1 to3

→  55

Input values AO←

Discrete Input block 1 to 2

→  56

Output values DI→

Discrete Output block 1 to3

→  57

Input values DO←

PROFIBUS DP

Defined order of modules
The measuring device works as a modular PROFIBUS slave. In contrast to a compact slave,a modular slave has a variable design and consists of several individual modules. The
device master file (GSD) contains a description of the individual modules (input and output
data) along with their individual properties.
The modules are permanently assigned to the slots, i.e. when configuring the modules, theorder and the arrangement of the modules must be respected.
Slot

Module

1…8

AI

9
10
11

TOTAL orSETTOT_TOTAL or
SETOT_MODETOT_TOTAL

Function block
Analog Input block 1 to 8
Totalizer block 1
Totalizer block 2
Totalizer block 3

12…14

AO

Analog Output block 1 to 3

15…16

DI

Discrete Input block 1 to 2

17…19

DO

Discrete Output block 1 to 3

To optimize the data throughput rate of the PROFIBUS network, it is advisable to only
configure modules that are processed in the PROFIBUS master system. If this results ingaps between the configured modules, these gaps must be assigned to the
EMPTY_MODULE.

9.4.2

Description of the modules

The data structure is described from the perspective of the PROFIBUS master:• Input data: Are sent from the measuring device to the PROFIBUS master.
• Output data: Are sent from the PROFIBUS master to the measuring device.

52

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

System integration

AI module (Analog Input)
Transmit an input variable from the measuring device to the PROFIBUS master (Class 1).
The selected input variable including its status is cyclically transmitted to the PROFIBUS
master (Class 1) via the AI module. The input variable is depicted in the first four bytes in
the form of a floating point number as per the IEEE 754 standard. The fifth byte contains
standardized status information pertaining to the input variable.
Eight Analog Input blocks are available (slot 1 to 8).
Selection: input variable
The input variable can be specified using the CHANNEL parameter.
CHANNEL

1)
2)

Input variable

32961

Mass flow

33122

Volume flow

33093

Corrected volume flow

708

Flow velocity

32850

Density

33092

Reference density

33101

Temperature

1042

Electronics temperature

901

Target mass flow 1)

793

Carrier mass flow 1)

794

Concentration 1)

263

Carrier tube temperature 2)

Only available with the Concentration application package
Only available with the Heartbeat Verification application package

Factory setting
Function block

Factory setting

AI 1

Mass flow

AI 2

DensityAI 3

Temperature

AI 4

Volume flow

AI 5

Corrected volume flow

AI 6

Reference density

AI 7

Mass flow

AI 8

Mass flow

Data structure
Input data of Analog Input
Byte 1

Byte 2Byte 3

Byte 4

Measured value: floating point number (IEEE 754)

Endress+Hauser

Byte 5
Status53

System integration

Proline Promass P 100 PROFIBUS DPTOTAL module
Transmit a totalizer value from the measuring device to the PROFIBUS master (Class 1).
A selected totalizer value, along with the status, is cyclically transmitted to a PROFIBUS
Master (Class 1) via the TOTAL module. The totalizer value is depicted in the first four
bytes in the form of a floating point number as per the IEEE 754 standard. The fifth byte
contains standardized status information pertaining to the totalizer value.
Three Totalizer blocks are available (slot 9 to 11).
Selection: totalizer value
The totalizer value can be specified using the CHANNEL parameter.
CHANNEL

1)

Input variable

32961

Mass flow

33122

Volume flow

33093

Corrected volume flow

901

Target fluid mass flow 1)

793

Carrier mass flow 1)

Only available with the "Concentration" application package

Factory setting
Function block

Factory setting: TOTAL

Totalizer 1, 2 and 3

Mass flow

Data structure
Input data of TOTAL
Byte 1

Byte 2Byte 3

Byte 4

Measured value: floating point number (IEEE 754)

Byte 5
StatusSETTOT_TOTAL module
The module combination consists of the SET_TOT and TOTAL functions:• SETTOT: Control the totalizers via the PROFIBUS master.
• TOTAL: Transmit totalizer value incl. status to PROFIBUS master.
Three Totalizer blocks are available (slot 9 to 11).
Selection: control totalizer
Value SETTOT

Control totalizer

0

Totalize

1

Resetting2

Adopt totalizer initial setting

Factory setting
Function block
Totalizer 1, 2 and 3

54

Factory setting: Value SETTOT (meaning)
0 (totalizing)

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

System integration

Data structure
Output data of SETTOT
Byte 1
Control variable 1

Input data of TOTAL
Byte 1

Byte 2Byte 3

Byte 4

Measured value: floating point number (IEEE 754)

Byte 5
StatusSETTOT_MODETOT_TOTAL module
The module combination consists of the SETTOT, MODETOT and TOTAL functions:• SETTOT: Control the totalizers via the PROFIBUS master.
• MODETOT: Configure the totalizers via the PROFIBUS master.
• TOTAL: Transmit totalizer value, along with the status, to the PROFIBUS master.
Three Totalizer blocks are available (slot 9 to 11).
Selection: totalizer configuration
MODETOT value

Totalizer configuration

0

Balancing

1

Balance the positive flow

2

Balance the negative flow

3

Stop totalizing

Factory setting
Function block
Totalizer 1, 2 and 3

Factory setting: Value MODETOT (meaning)
0 (balancing)

Data structure
Output data of SETTOT and MODETOT
Byte 1

Byte 2Control variable 1: SETTOT

Control variable 2: MODETOT

Input data of TOTAL
Byte 1

Byte 2Byte 3

Byte 4

Measured value: floating point number (IEEE 754)

Byte 5
Status

AO module (Analog Output)
Transmit a compensation value from the PROFIBUS master (class 1) to the measuring
device.
A compensation value, including the status, is cyclically transmitted from the PROFIBUS
master (class 1) to the measuring device via the AO module. The compensation value isdepicted in the first four bytes in the form of a floating point number as per the IEEE 754

Endress+Hauser

55

System integration

Proline Promass P 100 PROFIBUS DP

standard. The fifth byte contains standardized status information pertaining to thecompensation value.
Three Analog Output blocks are available (slot 12 to 14).
Assigned compensation values
A compensation value is permanently assigned to the individual Analog Output blocks.

1)

CHANNEL

Function block

Compensation value

306

AO 1

External pressure 1)

307

AO 2

External temperature 1)

488

AO 3

External reference density

The compensation values must be transmitted to the device in the SI basic unit

The selection is made via: Expert → Sensor → External compensation
Data structure
Output data of Analog Output
Byte 1

Byte 2Byte 3

Byte 4

Measured value: floating point number (IEEE 754)
1)

Byte 5
Status 1)

Status coding

DI module (Discrete Input)
Transmit discrete input values from the measuring device to the PROFIBUS master (class1). Discrete input values are used by the measuring device to transmit the state of device
functions to the PROFIBUS master (class 1).
The DI module cyclically transmits the discrete input value, including the status, to the
PROFIBUS master (class 1). The discrete input value is depicted in the first byte. The
second byte contains standardized status information pertaining to the input value.
Two Discrete Input blocks are available (slot 15 to 16).
Selection: device function
The device function can be specified using the CHANNEL parameter.
CHANNEL1)

Device function

893

Status switch output

894

Empty pipe detection

895

Low flow cut off

1430

Status verification 1)

Factory setting: Status (meaning)• 0 (device function not active)• 1 (device function active)

Only available with the Heartbeat Verification application package

Factory setting
Function block

56

Factory setting

DI 1

Empty pipe detection

DI 2

Low flow cut off

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

System integration

Data structure
Input data of Discrete Input
Byte 1

Byte 2Discrete

Status

DO module (Discrete Output)
Transmit discrete output values from the PROFIBUS master (class 1) to the measuring
device. Discrete output values are used by the PROFIBUS master (class 1) to enable anddisable device functions.
The DO module cyclically transmits the discrete output value, including the status, to themeasuring device. The discrete output value is depicted in the first byte. The second byte
contains standardized status information pertaining to the output value.
Three Discrete Output blocks are available (slot 17 to 19).
Assigned device functions
A device function is permanently assigned to the individual Discrete Output blocks.

1)

CHANNEL

Function block

Device function

891

DO 1

Flow override

890

DO 2

Zero adjustment

1429

DO 3

Start verification 1)

Values: control (meaning)• 0 (disable device function)• 1 (enable device function)

Only available with the Heartbeat Verification application package

Data structure
Output data of Discrete Output
Byte 1

Byte 2Discrete

StatusEMPTY_MODULE moduleThis module is used to assign empty spaces arising from modules not being used in theslots .
The measuring device works as a modular PROFIBUS slave. In contrast to a compact slave,a modular PROFIBUS slave has a variable design and consists of several individual
modules. The GSD file contains a description of the individual modules along with theirindividual properties.
The modules are permanently assigned to the slots. When configuring the modules, it isabsolutely essential to observe the sequence/arrangement of the modules. Any gaps
between the configured modules must be filled with the EMPTY_MODULE.

Endress+Hauser

57Commissioning

Proline Promass P 100 PROFIBUS DP

10Commissioning

10.1

Post-installation and post-connection check

Before commissioning the device:‣ Make sure that the post-installation and post-connection checks have been performed
successfully.

• "Post-installation check" checklist →  25
• "Post-connection check" checklist →  33

10.2

Connecting via FieldCare• For FieldCare connection• For connecting via FieldCare →  46
• For the FieldCare →  47 user interface

10.3

Setting the operating language

Factory setting: English or ordered local language
The operating language can be set in FieldCare, DeviceCare or via the Web server:
Operation → Display language

10.4

Configuring the measuring device

The Setup menu with its submenus contains all the parameters needed for standard
operation.

 Setup58

Device tag→  59‣ System units→  59‣ Medium selection→  62‣ Communication→  63‣ Analog inputs→  64‣ Low flow cut off→  65‣ Partially filled pipe detection→  66‣ Advanced setup→  67

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

10.4.1

Commissioning

Defining the tag nameTo enable fast identification of the measuring point within the system, you can enter aunique designation using the Device tag parameter and thus change the factory setting.
Enter the tag name in the "FieldCare" operating tool →  47
Navigation"Setup" menu → Device tag
Parameter overview with brief description
Parameter
Device tag

Description
Enter the name for the measuring point.

10.4.2User entry
Max. 32 characters, such as letters, numbers
or special characters (e.g. @, %, /).

Setting the system units

In the System units submenu the units of all the measured values can be set.
The number of submenus and parameters can vary depending on the device version.
Certain submenus and parameters in these submenus are not described in the
Operation Instructions. Instead a description is provided in the Special Documentation
for the device (→ "Supplementary documentation" section).
Navigation"Setup" menu → Advanced setup → System units‣ System units

Endress+Hauser

Mass flow unit

→  60Mass unit→  60Volume flow unit

→  60Volume unit

→  60

Corrected volume flow unit

→  60

Corrected volume unit

→  60Density unit

→  60

Reference density unit

→  60Temperature unit

→  61Pressure unit

→  6159

Commissioning

Proline Promass P 100 PROFIBUS DP

Parameter overview with brief description
Parameter
Mass flow unit

Description
Select mass flow unit.

Selection

Factory setting

Unit choose list

Country-specific:• kg/h• lb/minEffectThe selected unit applies for:• Output• Low flow cut off• Simulation process variable
Mass unit

Select mass unit.

Unit choose list

Country-specific:
• kg
• lb

Volume flow unit

Select volume flow unit.

Unit choose list

Depends on country:• l/h• gal/min (us)EffectThe selected unit applies for:• Output• Low flow cut off• Simulation process variable
Volume unit

Select volume unit.

Unit choose list

Country-specific:• l (DN > 150 (6"): m³ option)• gal (us)Corrected volume flow unit

Select corrected volume flow unit.

Unit choose list

Country-specific:• Nl/h• Sft³/min

ResultThe selected unit applies for:
Corrected volume flow parameter(→  79)
Corrected volume unit

Select corrected volume unit.

Unit choose list

Country-specific:• Nl• Sft³

Density unit

Select density unit.

Unit choose list

Country-specific:• kg/l• lb/ft³

EffectThe selected unit applies for:• Output• Simulation process variable• Density adjustment (Expert menu)
Reference density unit

Select reference density unit.

Unit choose list

Country-dependent• kg/Nl• lb/Sft³

Density 2 unit

Select second density unit.

Unit choose list

Depends on country:• kg/l• lb/ft³

60

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Parameter
Temperature unit

Commissioning

Description
Select temperature unit.

Selection
Unit choose list

Country-specific:
• °C
• °F

Unit choose list

Country-specific:• bar a• psi aEffectThe selected unit applies for:• Electronic temperature parameter(6053)• Maximum value parameter (6051)• Minimum value parameter (6052)• Maximum value parameter (6108)• Minimum value parameter (6109)• Carrier pipe temperature parameter(6027)• Maximum value parameter (6029)• Minimum value parameter (6030)• Reference temperature parameter(1816)• Temperature parameter
Pressure unit

Select process pressure unit.
ResultThe unit is taken from:• Pressure value parameter (→  63)• External pressure parameter (→  63)• Pressure value

Endress+Hauser

Factory setting

61Commissioning

Proline Promass P 100 PROFIBUS DP

10.4.3

Selecting and setting the medium

The Select medium wizard submenu contains parameters that must be configured inorder to select and set the medium.
Navigation"Setup" menu → Medium selection‣ Medium selection
Select medium→  62

Select gas type→  62

Reference sound velocity→  63Temperature coefficient sound velocity→  63Pressure compensation→  63Pressure value→  63External pressure→  63

Parameter overview with brief description
Parameter

Prerequisite

Description

Select medium–

Select gas typeIn the Medium selection submenu, the Select measured gas type.
Gas option is selected.

62

Use this function to select the type of
medium: "Gas" or "Liquid". Select the"Other" option in exceptional cases inorder to enter the properties of themedium manually (e.g. for highly
compressive liquids such as sulfuricacid).

Selection / User entry• Liquid
• Gas

• Air
• Ammonia NH3• Argon Ar• Sulfur hexafluoride SF6• Oxygen O2• Ozone O3• Nitrogen oxide NOx• Nitrogen N2• Nitrous oxide N2O• Methane CH4• Hydrogen H2• Helium He• Hydrogen chloride HCl• Hydrogen sulfide H2S• Ethylene C2H4
• Carbon dioxide CO2• Carbon monoxide CO• Chlorine Cl2• Butane C4H10• Propane C3H8
• Propylene C3H6
• Ethane C2H6
• Others

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Parameter

Commissioning

Prerequisite

Description

Reference sound velocity

In the Select gas type parameter, theOthers option is selected.

Enter sound velocity of gas at 0 °C (32°F).

1 to 99 999.9999 m/s

Temperature coefficient soundvelocityIn the Select gas type parameter, theOthers option is selected.

Enter temperature coefficient for thegas sound velocity.

Positive floating-point
numberPressure compensation–

Select pressure compensation type.

• Off• Fixed value• External value

Pressure value

In the Pressure compensation
parameter, the Fixed value option or
the Current input 1…n option is
selected.

Enter process pressure to be used forpressure correction.

Positive floating-point
numberExternal pressure

In the Pressure compensation
parameter, the External value option is
selected.

10.4.4

Selection / User entry

Configuring communication interface

The Communication submenu guides you systematically through all the parameters thathave to be configured for selecting and setting the communication interface.
Navigation"Setup" menu → Communication‣ Communication→  63

Device address

Parameter overview with brief description
Parameter
Device address

Endress+Hauser

Description
Enter device address.

User entry
0 to 126

63

Commissioning

Proline Promass P 100 PROFIBUS DP

10.4.5

Configuration of the Analog Inputs

The Analog inputs submenu guides the user systematically to the individual Analog input
1 to n submenu. From here you get to the parameters of the individual analog input.
Navigation"Setup" menu → Analog inputs‣ Analog inputs‣ Analog input 1 to n
Channel→  64

PV filter time→  64

Fail safe type→  64

Fail safe value→  64

Parameter overview with brief description
Parameter

Prerequisite

Description

Selection / User entry

Channel–Select the process variable.

• Mass flow
• Volume flow
• Corrected volume flow
• Target mass flow *• Carrier mass flow *• Density• Reference density• Concentration *• Temperature• Carrier pipe
temperature *• Electronic temperature• Oscillation frequency 0• Oscillation amplitude 0 *• Frequency fluctuation 0• Oscillation damping 0• Tube damping
fluctuation 0• Signal asymmetry• Exciter current 0

PV filter time–

Specify the time to suppress signal
peaks. During the specified time theAnalog Input does not respond to an
erratic increase in the process variable.

Positive floating-point
numberFail safe type–

Select the failure mode.

• Fail safe value• Fallback value• Off

Fail safe value

In Fail safe type parameter, the Failsafe value option is selected.

Specify the values to be output when an Signed floating-point
error occurs.
number*64

Visibility depends on order options or device settings

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

10.4.6

Commissioning

Configuring the low flow cut off

The Low flow cut off submenu contains the parameters that must be set in order toconfigure the low flow cut off.
Navigation"Setup" menu → Low flow cut off‣ Low flow cut off
Assign process variable→  65On value low flow cutoff→  65

Off value low flow cutoff→  65Pressure shock suppression→  65

Parameter overview with brief description
Parameter

Prerequisite

Description

Selection / User
entry

Factory setting

Select process variable for low
flow cut off.

• Off
• Mass flow
• Volume flow
• Corrected volume
flow–

Positive floatingpoint number

Depends on country
and nominal
diameter

Assign process variable–

On value low flow cutoff

A process variable is selectedEnter on value for low flow cut
in the Assign process variable off.
parameter (→  65).

Off value low flow cutoff

A process variable is selectedEnter off value for low flow cut 0 to 100.0 %in the Assign process variable off.
parameter (→  65).

–

Pressure shock suppression

A process variable is selectedEnter time frame for signal
in the Assign process variable suppression (= active pressure
parameter (→  65).
shock suppression).

–Endress+Hauser

0 to 100 s

65Commissioning

Proline Promass P 100 PROFIBUS DP

10.4.7

Configuring partially filled pipe detection

The Partially filled pipe detection submenu contains parameters that have to be set for
configuring empty pipe detection.
Navigation"Setup" menu → Partially filled pipe detection‣ Partially filled pipe detection
Assign process variable→  66Low value partial filled pipe detection→  66

High value partial filled pipe detection→  66

Response time part. filled pipe detect.

→  66

Parameter overview with brief description
Parameter

Prerequisite

Description

Selection / User
entry

Select process variable forpartially filled pipe detection.

• Off• Density• Reference density

Factory setting

Assign process variable–

Low value partial filled pipe
detection

A process variable is selectedEnter lower limit value for
Signed floating-pointin the Assign process variable deactivating partialy filled pipe number
parameter (→  66).
detection.

Depends on country:• 200 kg/m3• 12.5 lb/ft3

High value partial filled pipe
detection

A process variable is selectedEnter upper limit value for
Signed floating-pointin the Assign process variable deactivating partialy filled pipe number
parameter (→  66).
detection.

Depends on country:• 6 000 kg/m3• 374.6 lb/ft3

Response time part. filled pipe
detect.

A process variable is selected
Use this function to enter the
0 to 100 sin the Assign process variable minimum time (hold time) the
parameter (→  66).
signal must be present before
diagnostic message S962 "Pipeonly partly filled" is triggered in
the event of a partially filled or
empty measuring pipe.

–

66

–

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

10.5

Commissioning

Advanced settings

The Advanced setup submenu with its submenus contains parameters for specificsettings.
The number of submenus can vary depending on the device version, e.g. viscosity isavailable only with the Promass I.
Navigation"Setup" menu → Advanced setup‣ Advanced setup
Enter access code

‣ Calculated values→  67‣ Sensor adjustment→  69‣ Totalizer 1 to n

→  73‣ Display‣ Viscosity‣ Concentration‣ Heartbeat setup‣ Administration

10.5.1→  74Using the parameter to enter the access code

Navigation"Setup" menu → Advanced setup
Parameter overview with brief description
Parameter
Enter access code

Description
Enter access code to disable write protection of parameters.

10.5.2User entry
Max. 16-digit character string comprisingnumbers, letters and special characters

Calculated process variables

The Calculated values submenu contains parameters for calculating the corrected volume
flow.

Endress+Hauser

67Commissioning

Proline Promass P 100 PROFIBUS DP

Navigation"Setup" menu → Advanced setup → Calculated values‣ Calculated values‣ Corrected volume flow calculation→  68"Corrected volume flow calculation" submenu
Navigation"Setup" menu → Advanced setup → Calculated values → Corrected volume flow calculation‣ Corrected volume flow calculation
Corrected volume flow calculation(1812)→  68External reference density (6198)→  68Fixed reference density (1814)→  68Reference temperature (1816)→  69Linear expansion coefficient (1817)→  69Square expansion coefficient (1818)→  69Parameter overview with brief description
Parameter

Prerequisite

Description

Selection / Userinterface / User
entry

Factory setting

Corrected volume flow calculation–

Select reference density forcalculating the correctedvolume flow.

• Fixed reference
density• Calculated
reference density• Reference density
by API table 53• External reference
density–

External reference density

In the Corrected volume flow
calculation parameter, theExternal reference density
option is selected.

Shows external reference
density.

Floating point
number with sign–

Fixed reference density

The Fixed reference density
option is selected in the
Corrected volume flow
calculation parameterparameter.

Enter fixed value for reference
density.

Positive floatingpoint number–

68

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Parameter

Commissioning

Prerequisite

Description

Reference temperature

The Calculated reference
density option is selected in
the Corrected volume flow
calculation parameterparameter.

Enter reference temperature
for calculating the reference
density.

–273.15 to 99 999 °C Country-specific:• +20 °C
• +68 °F

Linear expansion coefficient

The Calculated reference
density option is selected in
the Corrected volume flow
calculation parameterparameter.

Enter linear, medium-specificexpansion coefficient forcalculating the reference
density.

Signed floating-point
number–Square expansion coefficient

The Calculated reference
density option is selected in
the Corrected volume flow
calculation parameterparameter.

For media with a non-linear
expansion pattern: enter thequadratic, medium-specificexpansion coefficient forcalculating the reference
density.

Signed floating-point
number

–10.5.3

Selection / Userinterface / User
entry

Factory setting

Carrying out a sensor adjustment

The Sensor adjustment submenu contains parameters that pertain to the functionality ofthe sensor.
Navigation"Setup" menu → Advanced setup → Sensor adjustment‣ Sensor adjustment→  69Installation direction‣ Density adjustment‣ Zero verification‣ Zero adjustment
Parameter overview with brief description
Parameter
Installation direction

Description
Set sign of flow direction to match the direction of the arrow on
the sensor.

Selection• Flow in arrow direction• Flow against arrow direction

Density adjustment
With density adjustment, a high level of accuracy is achieved only at the point ofadjustment and at the relevant density and temperature. However, the accuracy of adensity adjustment is only ever as good as the quality of the reference measuring data
provided. Therefore it is not a substitute for special density calibration.

Endress+Hauser

69Commissioning

Proline Promass P 100 PROFIBUS DP

Performing density adjustment
Note the following before performing the adjustment:• A density adjustment only makes sense if there is little variation in the operating
conditions and the density adjustment is performed under the operating conditions.
• The density adjustment scales the internally computed density value with a userspecific slope and offset.
• A 1-point or 2-point density adjustment can be performed.
• For a 2-point density adjustment, there must be a difference of at least 0.2 kg/l
between the two target density values.
• The reference media must be gas-free or pressurized so that any gas they contain iscompressed.
• The reference density measurements must be performed at the same medium
temperature that prevails in the process, as otherwise the density adjustment will
not be accurate.
• The correction resulting from the density adjustment can be deleted with the
Restore original option.
"1 point adjustment" option
1.

In the Density adjustment mode parameter, select the 1 point adjustment option
and confirm.

2.

In the Density setpoint 1 parameter, enter the density value and confirm.
 In the Execute density adjustment parameter the following options are now
available:
Ok
Measure density 1 option
Restore original

3.

Select the Measure density 1 option and confirm.

4.

If 100% was reached in the Progress parameter on the display and the Ok option isdisplayed in the Execute density adjustment parameter, then confirm.
 In the Execute density adjustment parameter the following options are now
available:
Ok
CalculateCancel5.

Select the Calculate option and confirm.

If the adjustment was completed successfully, the Density adjustment factor parameterand the Density adjustment offset parameter and the values calculated for them areshown on the display.
"2 point adjustment" option

70

1.

In the Density adjustment mode parameter, select the 2 point adjustment option
and confirm.

2.

In the Density setpoint 1 parameter, enter the density value and confirm.

3.

In the Density setpoint 2 parameter, enter the density value and confirm.
 In the Execute density adjustment parameter the following options are now
available:
Ok
Measure density 1
Restore originalEndress+Hauser

Proline Promass P 100 PROFIBUS DP

Commissioning

4.

Select the Measure density 1 option and confirm.
 In the Execute density adjustment parameter the following options are now
available:
Ok
Measure density 2
Restore original

5.

Select the Measure density 2 option and confirm.
 In the Execute density adjustment parameter the following options are now
available:
Ok
CalculateCancel6.

Select the Calculate option and confirm.

If the Density adjust failure option is displayed in the Execute density adjustment
parameter, call up the options and select the Cancel option. The density adjustment iscanceled and can be repeated.
If the adjustment was completed successfully, the Density adjustment factor parameterand the Density adjustment offset parameter and the values calculated for them areshown on the display.
Navigation"Expert" menu → Sensor → Sensor adjustment → Density adjustment‣ Density adjustment
Density adjustment mode→  71

Density setpoint 1→  71

Density setpoint 2→  72

Execute density adjustment→  72

Progress→  72

Density adjustment factor→  72

Density adjustment offset→  72

Parameter overview with brief description
Parameter

Prerequisite

Description

Selection / User
entry / Userinterface

Density adjustment mode–

• 1 point adjustment –• 2 point adjustment

Density setpoint 1–

The entry depends on –the unit selected in
the Density unit
parameter (0555).

Endress+Hauser

Factory setting

71Commissioning

Proline Promass P 100 PROFIBUS DP

Parameter

Prerequisite

Description

Selection / User
entry / Userinterface

Factory setting

Density setpoint 2

In the Density adjustmentmode parameter, the 2 point
adjustment option is selected.

The entry depends on –the unit selected in
the Density unit
parameter (0555).

Execute density adjustment–• Cancel• Busy• Ok• Density adjust
failure• Measure density 1• Measure density 2• Calculate• Restore original–

Progress–

0 to 100 %–Density adjustment factor–

Signed floating-point
number–Density adjustment offset–

Signed floating-point
number–Shows the progress of theprocess.

Zero point verification and zero adjustment
All measuring devices are calibrated in accordance with state-of-the-art technology.
Calibration takes place under reference conditions →  133. Therefore, a zero adjustmentin the field is generally not required.
Experience shows that zero adjustment is advisable only in special cases:• To achieve maximum measuring accuracy even with low flow rates.
• Under extreme process or operating conditions (e.g. very high process temperatures orvery high-viscosity fluids).
To get a representative zero point, ensure that:• any flow in the device is prevented during the adjustment• the process conditions (e.g. pressure, temperature) are stable and representative
Zero point verification and zero adjustment cannot be performed if the following process
conditions are present:• Gas pockets
Ensure that the system has been sufficiently flushed with the medium. Repeat flushing
can help to eliminate gas pockets• Thermal circulation
In the event of temperature differences (e.g. between the measuring tube inlet andoutlet section), induced flow can occur even if the valves are closed due to thermal
circulation in the device• Leaks at the valves
If the valves are not leak-tight, flow is not sufficiently prevented when determining thezero point
If these conditions cannot be avoided, it is advisable to keep the factory setting for the zero
point.

72

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Commissioning

Navigation"Setup" menu → Advanced setup → Sensor adjustment → Zero point adjustment‣ Zero point adjustment
Zero point adjustment control→  73

Progress→  73

Parameter overview with brief description
Parameter

Description

Selection / User interface

Factory setting

Zero point adjustment control

Start zero point adjustment.

• Cancel• Busy• Zero point adjust failure• Start–

Progress

Shows the progress of the process.

0 to 100 %–10.5.4

Configuring the totalizer

In the "Totalizer 1 to n" submenu the individual totalizer can be configured.
Navigation"Setup" menu → Advanced setup → Totalizer 1 to n

‣ Totalizer 1 to n

Endress+Hauser

Assign process variable→  74

Unit totalizer→  74Control Totalizer 1 to n

→  74

Totalizer operation mode→  74

Failure mode→  7473

Commissioning

Proline Promass P 100 PROFIBUS DP

Parameter overview with brief description
Parameter

Prerequisite

Description

Selection

Factory setting

Assign process variable–

Select process variable fortotalizer.

• Mass flow
• Volume flow
• Corrected volume
flow• Target mass flow *• Carrier mass flow *–Unit totalizer

One of the following options isselected in the Assign process
variable parameter:• Mass flow
• Volume flow
• Corrected volume flow
• Target mass flow *• Carrier mass flow *Select the unit for the process
variable of the totalizer.

Unit choose list

Country-specific:
• kg
• lb

Control Totalizer 1 to n

One of the following options isselected in the Assign process
variable parameter:• Mass flow
• Volume flow
• Corrected volume flow
• Target mass flow *• Carrier mass flow *Control the totalizer value.

• Totalize• Reset + hold
• Preset + hold–Totalizer operation mode

In the Assign process variable Select totalizer calculation
parameter, one of themode.
following options is selected:• Mass flow
• Volume flow
• Corrected volume flow
• Target mass flow *• Carrier mass flow *• Net flow total• Forward flow total• Reverse flow total• Last valid value–

Failure mode

In the Assign process variable Define the totalizer behavior in • Stop
parameter, one of thethe event of a device alarm.
• Actual value
following options is selected:• Last valid value• Mass flow
• Volume flow
• Corrected volume flow
• Target mass flow *• Carrier mass flow **–Visibility depends on order options or device settings

10.5.5Using parameters for device administration

The Administration submenu systematically guides the user through all the parametersthat can be used for device administration purposes.
Navigation"Setup" menu → Advanced setup → Administration‣ Administration

74

Define access code→  75

Device reset→  75Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Commissioning

Parameter overview with brief description
Parameter

Description

User entry / Selection

Define access code

Define release code for write access to parameters.

0 to 9 999

Device reset

Reset the device configuration - either entirely or in part - to adefined state.

• Cancel• To delivery settings• Restart device

10.6

Simulation

Via the Simulation submenu, it is possible to simulate various process variables in theprocess and the device alarm mode and verify downstream signal chains (switching valvesor closed-control loops). The simulation can be performed without a real measurement (noflow of medium through the device).
Navigation"Diagnostics" menu → Simulation‣ Simulation
Assign simulation process variable→  75

Value process variable→  75

Simulation device alarm→  75

Simulation diagnostic event→  76

Parameter overview with brief description
Parameter

Prerequisite

Description

Selection / User entry

Assign simulation process variable–

Select a process variable for thesimulation process that is activated.

• Off• Mass flow
• Volume flow
• Corrected volume flow
• Density• Reference density• Temperature• Concentration *• Target mass flow *• Carrier mass flow *Value process variable

A process variable is selected in the
Assign simulation process variable
parameter (→  75).

Enter the simulation value for theselected process variable.

Depends on the process
variable selected

Simulation device alarm–

Switch the device alarm on and off.

• Off
• On

Endress+Hauser

75Commissioning

Proline Promass P 100 PROFIBUS DP

Parameter

Prerequisite

Description

Selection / User entry

Diagnostic event category–

Select a diagnostic event category.

• Sensor• Electronics• Configuration• Process

Simulation diagnostic event–

Select a diagnostic event for thesimulation process that is activated.

• Off• Diagnostic event picklist(depends on the category
selected)*

Visibility depends on order options or device settings

10.7

Protecting settings from unauthorized access

The following options exist for protecting the configuration of the measuring device fromunintentional modification after commissioning:• Write protection via access code for Web browser →  76
• Write protection via write protection switch →  77

10.7.1

Write protection via access code

With the customer-specific access code, access to the measuring device via the Webbrowser is protected, as are the parameters for the measuring device configuration.
Navigation"Setup" menu → Advanced setup → Administration → Define access code

‣ Administration
Define access code→  75

Device reset→  75

Defining the access code via the Web browser
1.

Navigate to the Define access code parameter.

2.

Define a max. 16-digit numeric code as an access code.

3.

Enter the access code again in the to confirm the code.
 The Web browser switches to the login page.
If no action is performed for 10 minutes, the Web browser automatically returns to
the login page.
• If parameter write protection is activated via an access code, it can also only be
deactivated via this access code .
• The user role with which the user is currently logged on via Web browser isindicated by the Access status tooling parameter. Navigation path: Operation→ Access status tooling76

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

10.7.2

Commissioning

Write protection via write protection switch

The write protection switch makes it possible to block write access to the entire operating
menu with the exception of the following parameters:• External pressure• External temperature• Reference density• All parameters for configuring the totalizer
The parameter values are now read only and cannot be edited any more:• Via service interface (CDI-RJ45)• Via PROFIBUS DP
1.

Depending on the housing version, loosen the securing clamp or fixing screw of the
housing cover.

2.

Depending on the housing version, unscrew or open the housing cover anddisconnect the local display from the main electronics module where necessary→  143.

3.
1

2

2

3

4

4

8

5

16

PROFIBUS
address

OFF ON

1

6

32

7

64

8

- Software addressing

9- Write protection

10- Not used

A0021262

Setting the write protection switch on the main electronics module to the On position
enables hardware write protection. Setting the write protection switch on the mainelectronics module to the Off position (factory setting) disables hardware write
protection.
 If hardware write protection is enabled: the Locking status parameter displays
the Hardware locked option ; if disabled, the Locking status parameter does not
display any option .
4.

Endress+Hauser

Reverse the removal procedure to reassemble the transmitter.

77

Operation

Proline Promass P 100 PROFIBUS DP

11

Operation

11.1

Reading the device locking status

Device active write protection: Locking status parameter
Navigation"Operation" menu → Locking status
Function scope of "Locking status" parameter
Options

Description

Hardware locked

The write protection switch (DIP switch) for hardware locking is activated on the
I/O electronic module. This prevents write access to the parameters .

Temporarily locked

Write access to the parameters is temporarily locked on account of internal
processes running in the device (e.g. data upload/download, reset etc.). Once theinternal processing has been completed, the parameters can be changed once
again.

11.2

Adjusting the operating language

Detailed information:• To configure the operating language →  58
• For information on the operating languages supported by the measuring device→  144

11.3

Configuring the display

Detailed information:
On the advanced settings for the local display

11.4

Reading measured values

With the Measured values submenu, it is possible to read all the measured values.
Navigation"Diagnostics" menu → Measured values‣ Measured values

11.4.1‣ Process variables→  78‣ Totalizer 1 to n

→  80"Measured variables" submenu

The Process variables submenu contains all the parameters needed to display the current
measured values for each process variable.

78

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Operation

Navigation"Diagnostics" menu → Measured values → Measured variables‣ Measured variables
Mass flow→  79Volume flow

→  79Corrected volume flow

→  79Density→  80

Reference density→  80Temperature→  80Pressure→  80Concentration→  80Target mass flow

→  80Carrier mass flow

→  80Target corrected volume flow

→  80Carrier corrected volume flow

→  80Target volume flow

→  80Carrier volume flow

→  80Parameter overview with brief description
Parameter
Mass flow

Prerequisite–

Description
Displays the mass flow that is currently
measured.

User interface
Signed floating-point
numberDependency
The unit is taken from: Mass flow unit
parameter (→  60)
Volume flow–

Displays the volume flow that iscurrently calculated.

Signed floating-point
numberDependency
The unit is taken from the Volume flow
unit parameter (→  60).
Corrected volume flow–

Displays the corrected volume flow thatis currently calculated.

Signed floating-point
numberDependency
The unit is taken from: Correctedvolume flow unit parameter (→  60)

Endress+Hauser

79

Operation

Proline Promass P 100 PROFIBUS DP

Parameter
Density

Prerequisite–

Description
Shows the density currently measured.
Dependency
The unit is taken from the Density unit
parameter (→  60).

Reference density–

Displays the reference density that iscurrently calculated.

User interface
Signed floating-point
numberSigned floating-point
numberDependency
The unit is taken from: Reference
density unit parameter (→  60)
Temperature–

Shows the medium temperature
currently measured.

Signed floating-point
numberDependency
The unit is taken from: Temperatureunit parameter (→  61)
Pressure value–

Displays either a fixed or externalpressure value.

Signed floating-point
numberDependency
The unit is taken from the Pressure
unit parameter (→  61).
Concentration

For the following order code:
Order code for "Application package",
option ED "Concentration"


Target mass flow

With the following conditions:
Order code for "Application package",
option ED "Concentration"


Carrier mass flow

The software options currently
enabled are displayed in the
Software option overview
parameter.

The software options currently
enabled are displayed in the
Software option overview
parameter.

With the following conditions:
Order code for "Application package",
option ED "Concentration"



The software options currently
enabled are displayed in the
Software option overview
parameter.

Displays the concentration that iscurrently calculated.

Signed floating-point
numberDependency
The unit is taken from theConcentration unit parameter.

Displays the mass flow that is currently
measured for the target medium.

Signed floating-point
numberDependency
The unit is taken from the Mass flow
unit parameter (→  60).

Displays the mass flow of the carrier
medium that is currently measured.

Signed floating-point
numberDependency
The unit is taken from: Mass flow unit
parameter (→  60)

Target corrected volume flow–

Signed floating-point
numberCarrier corrected volume flow–

Signed floating-point
numberTarget volume flow–

Signed floating-point
numberCarrier volume flow–

Signed floating-point
number

11.4.2"Totalizer" submenu

The Totalizer submenu contains all the parameters needed to display the current
measured values for every totalizer.

80

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Operation

Navigation"Diagnostics" menu → Measured values → Totalizer‣ Totalizer
Totalizer value 1 to n

→  81

Totalizer overflow 1 to n

→  81

Parameter overview with brief description
Parameter

Prerequisite

Description

Totalizer value 1 to n

One of the following options is selectedin the Assign process variable
parameter of the Totalizer 1 to nsubmenu:• Volume flow
• Mass flow
• Corrected volume flow
• Target mass flow *• Carrier mass flow *Displays the current totalizer counter
reading.

Signed floating-point
numberTotalizer overflow 1 to n

One of the following options is selectedin the Assign process variable
parameter of the Totalizer 1 to nsubmenu:• Volume flow
• Mass flow
• Corrected volume flow
• Target mass flow *• Carrier mass flow *Displays the current totalizer overflow.

Integer with sign*

User interface

Visibility depends on order options or device settings

11.5

Adapting the measuring device to the process
conditions

The following are available for this purpose:• Basic settings using theSetup menu (→  58)• Advanced settings using theAdvanced setup submenu (→  67)

11.6

Performing a totalizer reset

The totalizers are reset in the Operation submenu:
Control Totalizer 1 to nFunction scope of "Control Totalizer " parameter

Endress+Hauser

Options

Description

Totalize

The totalizer is started.

Reset + hold

The totaling process is stopped and the totalizer is reset to 0.

Preset + hold

The totaling process is stopped and the totalizer is set to its defined start value
from the Preset value 1 to n parameter.

Stop totalizing option

Totalizing is stopped.

81

Operation

Proline Promass P 100 PROFIBUS DP

Navigation"Operation" menu → Totalizer handling‣ Totalizer handlingControl Totalizer 1 to n

→  82Preset value 1 to n

→  82

Reset all totalizers→  82

Parameter overview with brief description
Parameter

Prerequisite

Description

Selection / User entry

Control Totalizer 1 to n

One of the following options is selectedin the Assign process variable
parameter:• Mass flow
• Volume flow
• Corrected volume flow
• Target mass flow *• Carrier mass flow *Control the totalizer value.

• Totalize• Reset + hold
• Preset + holdPreset value 1 to n–Specify start value for totalizer.

Signed floating-point
numberReset all totalizers–

Reset all totalizers to 0 and start.

• Cancel• Reset + totalize*

82

Visibility depends on order options or device settings

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Diagnostics and troubleshooting

12

Diagnostics and troubleshooting

12.1

General troubleshooting

For local display
ProblemPossible causes

Remedial action

Local display is dark, but signal
output is within the valid range

The cable of the display module isnot plugged in correctly.

Insert the plug correctly into themain electronics module anddisplay module.

Local display dark and no output
signals

Supply voltage does not match thevoltage specified on the nameplate.

Apply the correct supply voltage→  29.

Local display dark and no output
signals

Supply voltage has incorrect
polarity.

Correct the polarity.

Local display dark and no output
signals

No contact between connecting
cables and terminals.

Ensure electrical contact between
the cable and the terminal.

Local display dark and no output
signals• Terminals are not plugged into
the I/O electronics module
correctly.
•

Check terminals.

Local display dark and no output
signals• I/O electronics module isdefective.
•

Order spare part →  123.

Local display is not readable, but
signal output is within the valid
range

Display is set too bright or too dark. • Set the display brighter bysimultaneously pressing  + .
• Set the display darker bysimultaneously pressing  + .

Local display is dark, but signal
output is within the valid range

Display module is defective.

Order spare part →  123.

Backlighting of local display is red

Diagnostic event with "Alarm"
diagnostic behavior has occurred.

Take remedial measures →  90

Message on local display:"Communication Error""Check Electronics"

Communication between thedisplay module and the electronics
is interrupted.

• Check the cable and theconnector between the mainelectronics module and display
module.
• Order spare part →  123.

Possible causes

Remedial action

For output signals
Error
Green power LED on the mainelectronics module of thetransmitter is dark

Supply voltage does not match thevalue indicated on the nameplate.

Apply the correct supply voltage→  29.

Device measures incorrectly.

Configuration error or device isoperated outside the application.

1. Check and correct parameter
configuration.
2. Observe limit values specified inthe "Technical Data".

For access
ProblemPossible causes

RemedyNo write access to parameters.

Hardware write protection is enabled.

Set the write protection switch on the mainelectronics module to the OFF position →  77.

No connection via PROFIBUS DP.

PROFIBUS DP bus cable is connected incorrectly.

Check the terminal assignment →  27.

No connection via PROFIBUS DP.

Device plug is not properly connected.

Check the pin assignment of the device plugs .

Endress+Hauser

83

Diagnostics and troubleshooting

Proline Promass P 100 PROFIBUS DP

ProblemPossible causes

RemedyNo connection via PROFIBUS DP.

PROFIBUS DP cable is incorrectly terminated.

Check the terminating resistor →  32.

No connection to web server.

Web server is disabled.

Using the "FieldCare" or "DeviceCare" operating
tool, check whether the Web server of themeasuring device is enabled, and enable it ifnecessary→  44.

Incorrect settings for the Ethernet interface ofthe computer.

1. Check the properties of the Internet protocol(TCP/IP) →  41.
2. Check the network settings with the IT
manager.

No connection to web server.

Incorrect IP address.

Check the IP address: 192.168.1.212→  41→  41

Web browser is frozen and operation no longer
possible.

Data transfer is active.

Wait until data transfer or current action isfinished.

Connection lost

1. Check cable connection and power supply.
2. Refresh the Web browser and restart if
necessary.

Content of web browser is incomplete or difficult Not using optimum version of Web server.
to read.

1. Use the correct Web browser version→  40.
2. Clear the Web browser cache and restart the
Web browser.

Unsuitable view settings.

Change the font size/display ratio of the Webbrowser.

No or incomplete display of contents in the webbrowser.

• JavaScript is not enabled• JavaScript cannot be enabled

1. Enable JavaScript.
2. Enter http://XXX.XXX.X.X.XX/servlet/
basic.html as the IP address.

Operation with FieldCare or DeviceCare is notpossible via CDI-RJ45 service interface (port
8000).

Firewall of computer or network is preventing
communication.

Depending on the settings of the firewall usedon the computer or in the network, the firewall
must be adapted or disabled to allow FieldCare/
DeviceCare access.

Flashing of firmware with FieldCare or
DeviceCare via CDI-RJ45 service interface (viaport 8000 or TFTP ports) is not possible.

Firewall of computer or network is preventing
communication.

Depending on the settings of the firewall usedon the computer or in the network, the firewall
must be adapted or disabled to allow FieldCare/
DeviceCare access.

12.2

Diagnostic information via light emitting diodes

12.2.1

Transmitter

Different LEDs in the transmitter provide information on the device status.
LED
Supply voltage

Alarm

Communication

84

Color

Meaning

Off

Supply voltage is off or too low

GreenSupply voltage is ok

OffDevice status is ok

Flashing red

A device error of diagnostic behavior "Warning" has occurredRed• A device error of diagnostic behavior "Alarm" has occurred• Boot loader is active

Flashing white

PROFIBUS DP communication is active

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Diagnostics and troubleshooting

12.3

Diagnostic information in the Web browser

12.3.1

Diagnostic options

Any faults detected by the measuring device are displayed in the Web browser on thehome page once the user has logged on.
1

2
A0032880

1
2

Status area with status signal
Diagnostics information →  85 and remedial measures with service ID

In addition, diagnostic events which have occurred can be shown in the Diagnostics
menu:• Via parameter →  115• Via submenu →  116
Status signals
The status signals provide information on the state and reliability of the device by
categorizing the cause of the diagnostic information (diagnostic event).
SymbolMeaning
FailureA device error has occurred. The measured value is no longer valid.
Function check
The device is in the service mode (during a simulation, for example).
Out of specification
The device is being operated:
Outside its technical specification limits (e.g. outside the process temperature range)
Maintenance required
Maintenance is required. The measured value is still valid.

The status signals are categorized in accordance with VDI/VDE 2650 and NAMUR
Recommendation NE 107.
Diagnostic information
The fault can be identified using the diagnostic information. The short text helps you byproviding information about the fault.

Endress+Hauser

85

Diagnostics and troubleshooting

Proline Promass P 100 PROFIBUS DP

Diagnostic information
Diagnostic code
Status signal

Diagnostic number

Short text

↓

↓

↓

842

Process limit

ExampleA0013958

3-digit number

12.3.2

Calling up remedy information

Remedy information is provided for every diagnostic event to ensure that problems can berectified quickly. These measures are displayed in red along with the diagnostic event andthe related diagnostic information.

12.4

Diagnostic information in FieldCare or DeviceCare

12.4.1

Diagnostic options

Any faults detected by the measuring device are displayed on the home page of theoperating tool once the connection has been established.
1
Xxxxxx/…/…/
Device name: Xxxxxxx

Mass flow:

12.34

kg/h

Device tag:

Volume flow:

12.34

m³/h

Xxxxxxx

Status signal:

XxxxxxDiagnostics 1:

P

Function check (C)C485 Simu...

P

Remedy information:

Deactivate...

P

Access status tooling:

Mainenance+Operation

+

Setup

+

Diagnostics+

ExpertFailure (F)2Function check (C)Diagnostics 1:

C485 Simulation measured vari...

Remedy information:

Deactivate Simulation (Service...

Out of spezification (S)3Maintenance required (M)A0021799-EN

1
2
3

Status area with status signal
Diagnostics information →  85
Remedial measures with service ID

In addition, diagnostic events which have occurred can be shown in the Diagnostics
menu:• Via parameter →  115• Via submenu →  116

86

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Diagnostics and troubleshooting

Diagnostic information
The fault can be identified using the diagnostic information. The short text helps you byproviding information about the fault.
Diagnostic information
Diagnostic code
Status signal

Diagnostic number

Short text

↓

↓

↓

842

Process limit

ExampleA0013958

3-digit number

12.4.2

Calling up remedy information

Remedy information is provided for every diagnostic event to ensure that problems can berectified quickly:• On the home page
Remedy information is displayed in a separate field below the diagnostics information.
• In the Diagnostics menu
Remedy information can be called up in the working area of the user interface.
The user is in the Diagnostics menu.
1.

Call up the desired parameter.

2.

On the right in the working area, mouse over the parameter.
 A tool tip with remedy information for the diagnostic event appears.

12.5

Adapting the diagnostic information

12.5.1

Adapting the diagnostic behavior

Each item of diagnostic information is assigned a specific diagnostic behavior at thefactory. The user can change this assignment for specific diagnostic information in the
Diagnostic behavior submenu.
Expert → System → Diagnostic handling → Diagnostic behavior/ ../Diagn. behavior
Diagnostic no.442

0658-1
WarningDiagnostic no.443

A0019179-EN

Available diagnostic behaviors
The following diagnostic behaviors can be assigned:

Endress+Hauser

Diagnostic behavior

Description

Alarm

The device stops measurement. The totalizers assume the defined alarm
condition. A diagnostic message is generated.

Warning

The device continues to measure. Measured value output via PROFIBUS andtotalizers are not affected. A diagnostic message is generated.

87

Diagnostics and troubleshooting

Proline Promass P 100 PROFIBUS DP

Diagnostic behavior

Description

Logbook entry only

The device continues to measure. The diagnostic message is only displayed in
the Event logbook submenu (Event list submenu) and is not displayed inalternating sequence with the operational display.

Off

The diagnostic event is ignored, and no diagnostic message is generated orentered.

Displaying the measured value status
If the Analog Input, Digital Input and Totalizer function blocks are configured for cyclic
data transmission, the device status is coded as per PROFIBUS PA Profile 3.02
Specification and transmitted along with the measured value to the PROFIBUS Master(Class 1) via the coding byte (byte 5). The coding byte is split into three segments: Quality,
Quality Substatus and Limits.

Measuring value

Coding(hex)
Byte 5

Quality

Quality Substatus

Limits
A0032228-EN

 16

Structure of the coding byte

The content of the coding byte depends on the configured failure mode in the individualfunction block. Depending on which failure mode has been configured, status informationin accordance with PROFINET PA Profile Specification 4 is transmitted to the PROFIBUS
master (Class 1) via the coding byte status information.
Determining the measured value status and device status via the diagnostic behavior
When the diagnostic behavior is assigned, this also changes the measured value status anddevice status for the diagnostic information. The measured value status and device statusdepend on the choice of diagnostic behavior and on the group in which the diagnostic
information is located.
The diagnostic information is grouped as follows:• Diagnostic information pertaining to the sensor: diagnostic number 000 to 199→  89• Diagnostic information pertaining to the electronics: diagnostic number 200 to 399→  89• Diagnostic information pertaining to the configuration: diagnostic number 400 to 599→  89• Diagnostic information pertaining to the process: diagnostic number 800 to 999→  90Depending on the group in which the diagnostic information is located, the following
measured value status and device status are firmly assigned to the particular diagnostic
behavior:

88

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Diagnostics and troubleshooting

Diagnostic information pertaining to the sensor: diagnostic number 000 to 199
Measured value status (fixed assignment)

Diagnostic behavior(configurable)

Quality

Quality
Substatus

Coding(hex)

Category(NE107)

Alarm

BAD

Maintenancealarm

0x24 to 0x27

F
(Failure)

Maintenancealarm

Warning

GOOD

Maintenancedemanded

0xA8 to 0xAB

M
(Maintenance)

MaintenancedemandedGOOD

ok0x80 to 0x8E––

Logbook entry only
Off

Device diagnosis(fixed assignment)

Diagnostic information pertaining to the electronics: diagnostic number 200 to 399
Diagnostic number 200 to 301, 303 to 399
Diagnostic behavior(configurable)
Alarm
Warning
Logbook entry onlyOff

Measured value status (fixed assignment)

Device diagnostics(fixed assignment)

Quality

Quality
Substatus

Coding(hex)

Category(NE107)BAD

Maintenancealarm

0x24 to 0x27

F
(Failure)

Maintenance
alarm

GOOD

ok

0x80 to 0x8E––

Diagnostic information 302
Measured value status (fixed assignment)

Diagnostic behavior(configurable)

Quality

Quality
Substatus

Coding(hex)

Category(NE107)

Device diagnostics(fixed assignment)

Alarm

BAD

Function check,
local override

0x3C to 0x3F

C

Function check

Warning

GOOD

Function check

0xBC to 0xBF––

Data logging continues when Heartbeat Verification is started. The signal outputs andtotalizers are not affected.
• Signal status: Function check• Choice of diagnostic behavior: alarm or warning (factory setting)
When Heartbeat verification starts, data logging is interrupted, the last valid measured
value is output and the totalizers are stopped.
Diagnostic information pertaining to the configuration: diagnostic number 400 to 599
Diagnostic behavior(configurable)
Alarm
Logbook entry only
Off
Logbook entry only
Off

Endress+Hauser

Measured value status (fixed assignment)

Device diagnosis(fixed assignment)

Quality

Quality
Substatus

Coding(hex)

Category(NE107)BAD

Function
check

0x3C to 0x3F

C
(Check)Function
check

GOOD

Function
check

0xBC to 0xBF–

Function
check

GOOD

ok

0x80 to 0x8E

–

–

89

Diagnostics and troubleshooting

Proline Promass P 100 PROFIBUS DP

Diagnostic information pertaining to the process: diagnostic number 800 to 999
Measured value status (fixed assignment)

Diagnostic behavior(configurable)

Quality

Quality
Substatus

Coding(hex)

Category(NE107)

Alarm

BAD

Process
related0x28 to 0x2B

F
(Failure)

Invalid process
condition

Warning

UNCERTA
IN

Process
related0x78 to 0x7B

S
(Out ofspecification)

Invalid process
condition

GOOD

ok

0x80 to 0x8E––

Logbook entry only
Off

12.6

Device diagnosis(fixed assignment)

Overview of diagnostic information• The amount of diagnostic information and the number of measured variables
affected increase if the measuring device has one or more application packages.
• All of the measured variables affected in the entire Promass instrument family arealways listed under "Measured variables affected". The measured variables available
for the device in question depend on the device version. When assigning themeasured variables to the device functions, for example to the individual outputs, allof the measured variables available for the device version in question are available
for selection.
In the case of some items of diagnostic information, the diagnostic behavior can bechanged. Adapting the diagnostic information →  87

12.6.1
Diagnostic information
No.

Remedy instructions

Short text

022 Sensor temperature

90

Diagnostic of sensor

1. Change main electronic module
2. Change sensor

Status signal

F

Diagnostic behavior

Alarm

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Diagnostic information
No.

Diagnostics and troubleshooting

Remedy instructions

Short text

046 Sensor limit exceeded

1. Inspect sensor
2. Check process condition

Status signal

S

Diagnostic behavior

WarningDiagnostic information
No.

Remedy instructions

Short text

062 Sensor connection

1. Change main electronic module
2. Change sensor

Status signal

F

Diagnostic behavior

Alarm

Diagnostic information
No.

Remedy instructions

Short text

082 Data storage

1. Check module connections
2. Contact service

Status signal

F

Diagnostic behavior

Alarm

Endress+Hauser

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Kinematic viscosity• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Volume flow

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Volume flow

91

Diagnostics and troubleshooting

Diagnostic information
No.

Remedy instructions

Short text

083 Memory content

1. Restart device
2. Contact service

Status signal

F

Diagnostic behavior

Alarm

Diagnostic information
No.

Remedy instructions

Short text

140 Sensor signal

1. Check or change main electronics
2. Change sensor

Status signal

S

Diagnostic behavior

Alarm

Diagnostic information
No.

Remedy instructions

Short text

144 Measuring error too high92

Proline Promass P 100 PROFIBUS DP

1. Check or change sensor
2. Check process conditions

Status signal

F

Diagnostic behavior

Alarm

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Kinematic viscosity• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Kinematic viscosity• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Diagnostic information
No.

Diagnostics and troubleshooting

Remedy instructions

Short text

190 Special event 1

Contact service

Status signal

F

Diagnostic behavior

Alarm

Diagnostic information
No.

Remedy instructions

Short text

191 Special event 5

Contact service

Status signal

F

Diagnostic behavior

Alarm

Endress+Hauser

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow

93

Diagnostics and troubleshooting

Diagnostic information
No.

Remedy instructions

Short text

192 Special event 9

Contact service

Status signal

F

Diagnostic behavior

Alarm

12.6.2
Diagnostic information
No.

Remedy instructions

1. Restart device
2. Contact service

Status signal

F

Diagnostic behavior

Alarm

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Kinematic viscosity• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Volume flow

Diagnostic of electronic

Short text

201 Device failure

94

Proline Promass P 100 PROFIBUS DP

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Diagnostic information
No.

Diagnostics and troubleshooting

Remedy instructions

Short text

242 Software incompatible
Status signal

F

Diagnostic behavior

Alarm

Diagnostic information
No.

1. Check software
2. Flash or change main electronics
moduleRemedy instructions

Short text

252 Modules incompatible

1. Check electronic modules
2. Change electronic modules

Status signal

F

Diagnostic behavior

Alarm

Endress+Hauser

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow

95

Diagnostics and troubleshooting

Diagnostic information
No.

Remedy instructions

Short text

262 Module connection

1. Check module connections
2. Change main electronics

Status signal

F

Diagnostic behavior

Alarm

Diagnostic information
No.

Status signal

F

Diagnostic behavior

Alarm

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow

Remedy instructions

Influenced measured
variables

Change main electronic module• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow

Short text

270 Main electronic failure

96

Proline Promass P 100 PROFIBUS DP

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Diagnostic information
No.

Diagnostics and troubleshooting

Remedy instructions

Short text

271 Main electronic failure

1. Restart device
2. Change main electronic module

Status signal

F

Diagnostic behavior

Alarm

Diagnostic information
No.

Remedy instructions

Short text

272 Main electronic failure

1. Restart device
2. Contact service

Status signal

F

Diagnostic behavior

Alarm

Endress+Hauser

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow

97

Diagnostics and troubleshooting

Diagnostic information
No.

Remedy instructions

Short text

273 Main electronic failure

Change electronic

Status signal

F

Diagnostic behavior

Alarm

Diagnostic information
No.

Remedy instructions

Short text

274 Main electronic failure

Change electronic

Status signal

S

Diagnostic behavior

WarningDiagnostic information
No.

Remedy instructions

Short text

283 Memory content

98

Proline Promass P 100 PROFIBUS DP

1. Reset device
2. Contact service

Status signal

F

Diagnostic behavior

Alarm

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow

Influenced measured
variables• Mass flow
• Sensor integrity• Corrected volume flow
• Volume flow

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Diagnostic information
No.

Diagnostics and troubleshooting

Remedy instructions

Short text

311 Electronic failure

1. Reset device
2. Contact service

Status signal

F

Diagnostic behavior

Alarm

Diagnostic information
No.

Remedy instructions

Short text

311 Electronic failure

1. Do not reset device
2. Contact service

Status signal

M

Diagnostic behavior

WarningEndress+Hauser

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Volume flow

99Diagnostics and troubleshooting

Diagnostic information
No.

Remedy instructions

Short text

382 Data storage

1. Insert DAT module
2. Change DAT module

Status signal

F

Diagnostic behavior

Alarm

Diagnostic information
No.

Short text

383 Memory content

100

Proline Promass P 100 PROFIBUS DP

Status signal

F

Diagnostic behavior

Alarm

Remedy instructions

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow

Influenced measured
variables

1. Restart device• Carrier mass flow
2. Check or change DAT module 3. Contact • Concentration
service• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Diagnostic information
No.

Diagnostics and troubleshooting

Remedy instructions

Short text

390 Special event 2

Contact service

Status signal

F

Diagnostic behavior

Alarm

Diagnostic information
No.

Remedy instructions

Short text

391 Special event 6

Contact service

Status signal

F

Diagnostic behavior

Alarm

Endress+Hauser

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow101Diagnostics and troubleshooting

Diagnostic information
No.

Remedy instructions

Short text

392 Special event 10

Contact service

Status signal

F

Diagnostic behavior

Alarm

12.6.3
Diagnostic information
No.

Remedy instructions

1. Check connection
2. Retry data transfer

Status signal

F

Diagnostic behavior

Alarm

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Kinematic viscosity• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Volume flow

Diagnostic of configuration

Short text

410 Data transfer

102

Proline Promass P 100 PROFIBUS DP

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Volume flow

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Diagnostic information
No.

Diagnostics and troubleshooting

Remedy instructions

Influenced measured
variables

Up-/download active, please wait

• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Volume flow

Remedy instructions

Influenced measured
variables

Up-/download active, please wait

• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Volume flow

Short text

411 Up-/download active
Status signal

C

Diagnostic behavior

WarningDiagnostic information
No.

Short text

411 Up-/download active
Status signal

C

Diagnostic behavior

WarningEndress+Hauser

103Diagnostics and troubleshooting

Diagnostic information
No.

Remedy instructions

Short text

437 Configuration incompatible

1. Restart device
2. Contact service

Status signal

F

Diagnostic behavior

Alarm

Diagnostic information
No.

Status signal

M

Diagnostic behavior

Warning

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow

Remedy instructions

Influenced measured
variables

1. Check data set file
2. Check device configuration
3. Up- and download new configuration• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Volume flow

Short text

438 Dataset

104

Proline Promass P 100 PROFIBUS DP

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Diagnostic information
No.

Diagnostics and troubleshooting

Remedy instructions

Influenced measured
variables

Short text

453 Flow override

Deactivate flow override

Status signal

C

Diagnostic behavior

WarningDiagnostic information
No.

• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Volume flow

Remedy instructions

Influenced measured
variables

Short text

482 FB not Auto/Cas

Set Block in AUTO mode

Status signal

F

Diagnostic behavior

Alarm

Diagnostic information
No.

Remedy instructions

Short text

484 Simulation failure mode

Deactivate simulation

Status signal

C

Diagnostic behavior

Alarm

Endress+Hauser–

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Volume flow

105

Diagnostics and troubleshooting

Diagnostic information
No.

Remedy instructions

Influenced measured
variables

Short text

485 Simulation measured variable

Deactivate simulation

Status signal

C

Diagnostic behavior

WarningDiagnostic information
No.

• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Volume flow

Remedy instructions

Influenced measured
variables

Short text

495 Simulation diagnostic event

Deactivate simulation

Status signal

C

Diagnostic behavior

WarningDiagnostic information
No.

–

Remedy instructions

Influenced measured
variables

Short text

497 Simulation block output

Deactivate simulation

Status signal

C

Diagnostic behavior

WarningDiagnostic information
No.

–

Remedy instructions

Influenced measured
variables

Short text

537 Configuration

106

Proline Promass P 100 PROFIBUS DP

1. Check IP addresses in network
2. Change IP address

Status signal

F

Diagnostic behavior

Warning–Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Diagnostic information
No.

Diagnostics and troubleshooting

Remedy instructions

Short text

590 Special event 3

Contact service

Status signal

F

Diagnostic behavior

Alarm

Diagnostic information
No.

Remedy instructions

Short text

591 Special event 7

Contact service

Status signal

F

Diagnostic behavior

Alarm

Endress+Hauser

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow

107

Diagnostics and troubleshooting

Proline Promass P 100 PROFIBUS DP

Diagnostic information
No.

Short text

592 Special event 11

Contact service

Status signal

F

Diagnostic behavior

Alarm

12.6.4
Diagnostic information
No.

Remedy instructions

1. Check ambient temperature
2. Check process temperature

Status signal

S

Diagnostic behavior

WarningDiagnostic information
No.

Remedy instructions

Short text

825 Operating temperature

1. Check ambient temperature
2. Check process temperature

Status signal

S

Diagnostic behavior

Warning

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Kinematic viscosity• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Volume flow

Diagnostic of process

Short text

825 Operating temperature108

Remedy instructions

Influenced measured
variables
Volume flow

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Diagnostic information
No.

Diagnostics and troubleshooting

Remedy instructions

Short text

825 Operating temperature

1. Check ambient temperature
2. Check process temperature

Status signal

F

Diagnostic behavior

Alarm

Diagnostic information
No.

Remedy instructions

Short text

830 Sensor temperature too highReduce ambient temp. around the sensor
housing

Status signal

S

Diagnostic behavior

WarningDiagnostic information
No.

Remedy instructions

Short text

831 Sensor temperature too low

Increase ambient temp. around the sensor
housing

Status signal

S

Diagnostic behavior

WarningEndress+Hauser

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Kinematic viscosity• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Volume flow

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Kinematic viscosity• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Volume flow

109

Diagnostics and troubleshooting

Diagnostic information
No.

Remedy instructions

Short text

832 Electronic temperature too highStatus signal

S

Diagnostic behavior

WarningDiagnostic information
No.

Reduce ambient temperature

Remedy instructions

Short text

833 Electronic temperature too low

Increase ambient temperature

Status signal

S

Diagnostic behavior

WarningDiagnostic information
No.

Remedy instructions

Short text

834 Process temperature too high110

Proline Promass P 100 PROFIBUS DP

Reduce process temperature

Status signal

S

Diagnostic behavior

Warning

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Empty pipe detection
option• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temperature• Volume flow

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Volume flow

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Kinematic viscosity• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Volume flow

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Diagnostic information
No.

Diagnostics and troubleshooting

Remedy instructions

Short text

835 Process temperature too low

Increase process temperature

Status signal

S

Diagnostic behavior

WarningDiagnostic information
No.

Remedy instructions

Short text

842 Process limit

Low flow cut off active!
1. Check low flow cut off configuration

Status signal

S

Diagnostic behavior

WarningDiagnostic information
No.

Remedy instructions

Short text

843 Process limit

Check process conditions

Status signal

S

Diagnostic behavior

WarningEndress+Hauser

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Kinematic viscosity• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Volume flow

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Volume flow

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Volume flow

111

Diagnostics and troubleshooting

Diagnostic information
No.

Remedy instructions

Short text

862 Partly filled pipe

1. Check for gas in process
2. Adjust detection limits

Status signal

S

Diagnostic behavior

WarningDiagnostic information
No.

Remedy instructions

Short text

882 Input signal
Status signal

F

Diagnostic behavior

Alarm

Diagnostic information
No.

1. Check input configuration
2. Check external device or process
conditions

Remedy instructions

Short text

910 Tubes not oscillating

112

Proline Promass P 100 PROFIBUS DP

1. Check electronic
2. Inspect sensor

Status signal

F

Diagnostic behavior

Alarm

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Kinematic viscosity• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Volume flow

Influenced measured
variables• Density• Mass flow
• Reference density• Corrected volume flow
• Volume flow

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Empty pipe detection
option• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temperature• Volume flow

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Diagnostic information
No.

Diagnostics and troubleshooting

Remedy instructions

Short text

912 Medium inhomogeneous

1. Check process cond.
2. Increase system pressure

Status signal

S

Diagnostic behavior

WarningDiagnostic information
No.

Remedy instructions

Short text

912 Inhomogeneous

1. Check process cond.
2. Increase system pressure

Status signal

S

Diagnostic behavior

WarningDiagnostic information
No.

Remedy instructions

Short text

913 Medium unsuitable

1. Check process conditions
2. Check electronic modules or sensor

Status signal

S

Diagnostic behavior

WarningEndress+Hauser

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Volume flow

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Volume flow

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Kinematic viscosity• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Volume flow

113

Diagnostics and troubleshooting

Diagnostic information
No.

Remedy instructions

Influenced measured
variables

Short text

944 Monitoring failed

Check process conditions for Heartbeat
Monitoring

Status signal

S

Diagnostic behavior

WarningDiagnostic information
No.

• Carrier mass flow
• Concentration• Density• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temperature

Remedy instructions

Influenced measured
variables

Short text

948 Tube damping too highCheck process conditions

Status signal

S

Diagnostic behavior

WarningDiagnostic information
No.

Remedy instructions

Short text

990 Special event 4

114

Proline Promass P 100 PROFIBUS DP

Contact service

Status signal

F

Diagnostic behavior

Alarm

–

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Diagnostic information
No.

Diagnostics and troubleshooting

Remedy instructions

Short text

991 Special event 8

Contact service

Status signal

F

Diagnostic behavior

Alarm

Diagnostic information
No.

Remedy instructions

Short text

992 Special event 12

Contact service

Status signal

F

Diagnostic behavior

Alarm

12.7

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Empty pipe detection
option• Kinematic viscosity• Low flow cut off option• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Status• Volume flow

Influenced measured
variables• Carrier mass flow
• Concentration• Density• Dynamic viscosity• Kinematic viscosity• Mass flow
• Sensor integrity• Reference density• Corrected volume flow
• Target mass flow
• Temp. compensateddynamic viscosity• Temp. compensatedkinematic viscosity• Temperature• Volume flow

Pending diagnostic events

The Diagnostics menu allows the user to view the current diagnostic event and theprevious diagnostic event separately.
To call up the measures to rectify a diagnostic event:• Via web browser →  86
• Via "FieldCare" operating tool →  87
• Via "DeviceCare" operating tool →  87
Other pending diagnostic events can be displayed in the Diagnostic list submenu→  116

Endress+Hauser

115

Diagnostics and troubleshooting

Proline Promass P 100 PROFIBUS DP

Navigation"Diagnostics" menu

 Diagnostics
Actual diagnostics→  116

Previous diagnostics→  116

Operating time from restart→  116

Operating time→  116

Parameter overview with brief description
Parameter
Actual diagnostics

Prerequisite
A diagnostic event has occurred.

Description

User interface

Shows the current occured diagnostic
event along with its diagnostic
information.

Symbol for diagnostic
behavior, diagnostic codeand short message.

two or more messages occur
 Ifsimultaneously,
the message withthe highest priority is shown on
the display.
Previous diagnostics

Two diagnostic events have alreadyoccurred.

Shows the diagnostic event that
occurred prior to the current diagnostic
event along with its diagnostic
information.

Symbol for diagnostic
behavior, diagnostic codeand short message.

Operating time from restart–Shows the time the device has been in
operation since the last device restart.

Days (d), hours (h),minutes (m) and seconds(s)Operating time–

Indicates how long the device has beenin operation.

Days (d), hours (h),minutes (m) and seconds
(s)

12.8

Diagnostic list

Up to 5 currently pending diagnostic events can be displayed in the Diagnostic list
submenu along with the associated diagnostic information. If more than 5 diagnostic
events are pending, the events with the highest priority are shown on the display.
Navigation path
Diagnostics → Diagnostic list
To call up the measures to rectify a diagnostic event:• Via web browser →  86
• Via "FieldCare" operating tool →  87
• Via "DeviceCare" operating tool →  87

12.9

Event logbook

12.9.1

Reading out the event logbook

A chronological overview of the event messages that have occurred is provided in the
Events list submenu.
116

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Diagnostics and troubleshooting

Navigation path
Diagnostics menu → Event logbook submenu → Event list
The event history includes entries for:• Diagnostic events →  90
• Information events →  117
In addition to the operation time of its occurrence, each event is also assigned a symbolthat indicates whether the event has occurred or is ended:• Diagnostic event• : Occurrence of the event• : End of the event• Information event
: Occurrence of the event
To call up the measures to rectify a diagnostic event:• Via web browser →  86
• Via "FieldCare" operating tool →  87
• Via "DeviceCare" operating tool →  87
For filtering the displayed event messages →  117

12.9.2Filtering the event logbook

Using the Filter options parameter you can define which category of event message isdisplayed in the Events list submenu.
Navigation path
Diagnostics → Event logbook → Filter options
Filter categories• All• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• Information (I)12.9.3

Overview of information events

Unlike a diagnostic event, an information event is displayed in the event logbook only andnot in the diagnostic list.
Info number

Endress+Hauser

Info name

I1000--------(Device ok)

I1089

Power on

I1090

Configuration reset

I1091

Configuration changed

I1110

Write protection switch changed

I1111

Density adjust failure

I1137

Electronic changed

I1151

History reset

I1155

Reset electronic temperature

I1157

Memory error event list

I1185

Display backup done

I1186

Restore via display done

117

Diagnostics and troubleshooting

Proline Promass P 100 PROFIBUS DPInfo number

Info name

I1187

Settings downloaded with display

I1188

Display data cleared

I1189

Backup compared

I1209

Density adjustment ok

I1221

Zero point adjust failure

I1222

Zero point adjustment ok

I1256

Display: access status changed

I1264

Safety sequence aborted

I1335

Firmware changed

I1361

Wrong web server login

I1397

Fieldbus: access status changed

I1398

CDI: access status changed

I1444

Device verification passed

I1445

Device verification failed

I1446

Device verification active

I1447

Record application reference data

I1448

Application reference data recorded

I1449

Recording application ref. data failed

I1450

Monitoring off

I1451

Monitoring on

I1457

Failed:Measured error verification

I1459

Failed: I/O module verification

I1460

Failed: Sensor integrity verification

I1461

Failed: Sensor verification

I1462

Failed:Sensor electronic module verific.

12.10 Resetting the measuring device
The entire device configuration or some of the configuration can be reset to a defined statewith the Device reset parameter (→  75).

12.10.1 Function scope of "Device reset" parameter
Options

Description

CancelNo action is executed and the user exits the parameter.

To delivery settings

Every parameter for which a customer-specific default setting was ordered is resetto the customer-specific value. All other parameters are reset to the factory setting.

 This option is not visible if no customer-specific settings have been ordered.
Restart device

118

The restart resets every parameter with data stored in volatile memory (RAM) to
the factory setting (e.g. measured value data). The device configuration remains
unchanged.

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Diagnostics and troubleshooting

12.11 Device information
The Device information submenu contains all parameters that display different
information for device identification.
Navigation"Diagnostics" menu → Device information‣ Device information
Device tag→  119

Serial number→  119

Firmware version→  119

Device name→  120

Order code→  120

Extended order code 1→  120

Extended order code 2→  120

Extended order code 3→  120

ENP version→  120

PROFIBUS ident number→  120

Status PROFIBUS Master Config→  120

IP address
Subnet mask
Default gateway

Parameter overview with brief description
Parameter

Description

User interface

Factory setting

Device tag

Shows name of measuring point.

Max. 32 characters, such as
letters, numbers or special
characters (e.g. @, %, /).

Promass 100 DP

Serial number

Shows the serial number of the measuring
device.

Max. 11-digit character string
comprising letters andnumbers.

–

Firmware version

Shows the device firmware version installed.

Character string in the format
xx.yy.zz–

Endress+Hauser

119Diagnostics and troubleshooting

Parameter
Order code

Proline Promass P 100 PROFIBUS DP

Description
Shows the device order code.
The order code can be found on the nameplateof the sensor andUser interface

Factory setting

Character string composed ofletters, numbers and certain
punctuation marks (e.g. /).

–Max. 32 characters such as
letters or numbers.

–

Character string–

Character string–

Character string–transmitter in the "Order code" field.
Device nameShows the name of the transmitter.

Extended order code 1

The name can be found on the
nameplate of the transmitter.

Shows the 1st part of the extended order
code.
extended order code can also be Thefound on the nameplate of the sensor
and transmitter in the "Ext. ord. cd."
field.

Extended order code 2

Shows the 2nd part of the extended order
code.
extended order code can also be Thefound on the nameplate of the sensor
and transmitter in the "Ext. ord. cd."
field.

Extended order code 3

Shows the 3rd part of the extended order
code.
extended order code can also be Thefound on the nameplate of the sensor
and transmitter in the "Ext. ord. cd."
field.

ENP version

Shows the version of the electronic
nameplate (ENP).

Character string–

PROFIBUS ident number

Displays the PROFIBUS identification
number.

0 to FFFF

0x1561

Status PROFIBUS Master Config

Displays the status of the PROFIBUS Masterconfiguration.

• Active• Not active–

120

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Diagnostics and troubleshooting

12.12 Firmware history
Releasedate

Firmwareversion

Order codefor"Firmwareversion"FirmwarechangesDocumentation
type

Documentation

09.2013 01.00.00

Option 78

Original firmware

OperatingInstructions

BA01253D/06/EN/01.13

10.2014 01.01.zz

Option 69• Integration of
optional local
display• New unit "BeerBarrel (BBL)"• Simulation ofdiagnostic events

OperatingInstructions

BA01253D/06/EN/02.14

It is possible to flash the firmware to the current version or the previous version using
the service interface.
For the compatibility of the firmware version with the previous version, the installed
device description files and operating tools, observe the information about the device
in the "Manufacturer's information" document.
The manufacturer's information is available:• In the Download Area of the Endress+Hauser web site: www.endress.com →Downloads• Specify the following details:• Product root, e.g. 8E1B
The product root is the first part of the order code: see the nameplate on thedevice.
• Text search: Manufacturer's information• Media type: Documentation – Technical Documentation

Endress+Hauser

121

Maintenance

Proline Promass P 100 PROFIBUS DP

13

Maintenance

13.1

Maintenance tasks

No special maintenance work is required.

13.1.1

Exterior cleaning

When cleaning the exterior of measuring devices, always use cleaning agents that do notattack the surface of the housing or the seals.

13.1.2

Interior cleaning

Observe the following points for CIP and SIP cleaning:• Use only cleaning agents to which the process-wetted materials are adequately resistant.
• Observe the maximum permitted medium temperature for the measuring device→  137.
Observe the following point for cleaning with pigs:
Observe the inside diameter of the measuring tube and process connection.

13.2

Measuring and test equipment

Endress+Hauser offers a wide variety of measuring and test equipment, such as W@M or
device tests.
Your Endress+Hauser Sales Center can provide detailed information on the services.
List of some of the measuring and testing equipment: →  125→  126

13.3

Endress+Hauser services

Endress+Hauser offers a wide variety of services for maintenance such as recalibration,
maintenance service or device tests.
Your Endress+Hauser Sales Center can provide detailed information on the services.

122

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Repair14

Repair

14.1

General information

14.1.1

Repair and conversion concept

The Endress+Hauser repair and conversion concept provides for the following:• The measuring devices have a modular design.
• Spare parts are grouped into logical kits with the associated Installation Instructions.
• Repairs are carried out by Endress+Hauser Service or by appropriately trained customers.
• Certified devices can only be converted to other certified devices by Endress+Hauser
Service or at the factory.

14.1.2

Notes for repair and conversion

For repair and modification of a measuring device, observe the following notes:‣ Use only original Endress+Hauser spare parts.
‣ Carry out the repair according to the Installation Instructions.
‣ Observe the applicable standards, federal/national regulations, Ex documentation (XA)
and certificates.

‣ Document every repair and each conversion and enter them into the W@M life cycle
management database and Netilion Analytics.

14.2

Spare parts

Device Viewer (www.endress.com/deviceviewer):
All the spare parts for the measuring device, along with the order code, are listed here andcan be ordered. If available, users can also download the associated Installation
Instructions.
Measuring device serial number:• Is located on the nameplate of the device.
• Can be read out via the Serial number parameter (→  119) in the Device
information submenu.

14.3

Endress+Hauser services

Endress+Hauser offers a wide range of services.
Your Endress+Hauser Sales Center can provide detailed information on the services.

14.4

ReturnThe requirements for safe device return can vary depending on the device type andnational legislation.

Endress+Hauser

1.

Refer to the web page for information:
http://www.endress.com/support/return-material
 Select the region.

2.

Return the device if repairs or a factory calibration are required, or if the wrong
device was ordered or delivered.

123

Repair

Proline Promass P 100 PROFIBUS DP

14.5

Disposal

If required by the Directive 2012/19/EU on waste electrical and electronic equipment(WEEE), the product is marked with the depicted symbol in order to minimize thedisposal of WEEE as unsorted municipal waste. Do not dispose of products bearing thismarking as unsorted municipal waste. Instead, return them to the manufacturer for
disposal under the applicable conditions.

14.5.1
1.

Removing the measuring device

Switch off the device.

LWARNING
Danger to persons from process conditions!
‣ Beware of hazardous process conditions such as pressure in the measuring device, hightemperatures or aggressive media.
2.

Carry out the mounting and connection steps from the "Mounting the measuring
device" and "Connecting the measuring device" sections in reverse order. Observe thesafety instructions.

14.5.2

Disposing of the measuring device

LWARNING
Danger to personnel and environment from fluids that are hazardous to health.
‣ Ensure that the measuring device and all cavities are free of fluid residues that arehazardous to health or the environment, e.g. substances that have permeated into
crevices or diffused through plastic.
Observe the following notes during disposal:‣ Observe valid federal/national regulations.
‣ Ensure proper separation and reuse of the device components.

124

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

15

Accessories

Accessories

Various accessories, which can be ordered with the device or subsequently from Endress+Hauser, are available for the device. Detailed information on the order code in question isavailable from your local Endress+Hauser sales center or on the product page of theEndress+Hauser website: www.endress.com.

15.1

Device-specific accessories

15.1.1For the sensor

Accessories

Description

Heating jacket

Is used to stabilize the temperature of the fluids in the sensor. Water, water vaporand other non-corrosive liquids are permitted for use as fluids.

 If using oil as a heating medium, please consult with Endress+Hauser.
• If ordered together with the measuring device:
Order code for "Accessory enclosed"• Option RB "Heating jacket, G 1/2" female thread"• Option RC "Heating jacket, G 3/4" female thread"• Option RD "Heating jacket, NPT 1/2" female thread"• Option RE "Heating jacket, NPT 3/4" female thread"• If ordered subsequently:
Use the order code with the product root DK8003.

 Special Documentation SD02160D
15.2

Communication-specific accessories

Accessories

Description

Commubox FXA291

Connects Endress+Hauser field devices with a CDI interface (= Endress+Hauser
Common Data Interface) and the USB port of a computer or laptop.

 Technical Information TI405C/07
Fieldgate FXA42

Is used to transmit the measured values of connected 4 to 20 mA analog
measuring devices, as well as digital measuring devices
Technical Information TI01297S
 •• Operating
Instructions BA01778S• Product page: www.endress.com/fxa42

Field Xpert SMT50

The Field Xpert SMT70 tablet PC for device configuration enables mobile plant
asset management in non-hazardous areas. It is suitable for commissioning andmaintenance staff to manage field instruments with a digital communicationinterface and to record progress.
This tablet PC is designed as an all-in-one solution with a preinstalled driver libraryand is an easy-to-use, touch-sensitive tool which can be used to manage field
instruments throughout their entire life cycle.
Technical Information TI01342S
 •• Operating
Instructions BA01709S• Product page: www.endress.com/smt50

Endress+Hauser

125

Accessories

Proline Promass P 100 PROFIBUS DP

Field Xpert SMT70

The Field Xpert SMT70 tablet PC for device configuration enables mobile plant
asset management in hazardous and non-hazardous areas. It is suitable forcommissioning and maintenance staff to manage field instruments with a digitalcommunication interface and to record progress.
This tablet PC is designed as an all-in-one solution with a preinstalled driver libraryand is an easy-to-use, touch-sensitive tool which can be used to manage field
instruments throughout their entire life cycle.
Technical Information TI01342S
 •• Operating
Instructions BA01709S• Product page: www.endress.com/smt70

Field Xpert SMT77

The Field Xpert SMT77 tablet PC for device configuration enables mobile plant
asset management in areas categorized as Ex Zone 1.
Technical Information TI01418S
 •• Operating
Instructions BA01923S• Product page: www.endress.com/smt77

15.3

Service-specific accessories

Accessory

Description

Applicator

Software for selecting and sizing Endress+Hauser measuring devices:• Choice of measuring devices with industrial requirements• Calculation of all the necessary data for identifying the optimum flowmeter:
e.g. nominal diameter, pressure loss, flow velocity and accuracy.
• Graphic illustration of the calculation results• Determination of the partial order code, administration, documentation andaccess to all project-related data and parameters over the entire life cycle ofa project.
Applicator is available:• Via the Internet: https://portal.endress.com/webapp/applicator• As a downloadable DVD for local PC installation.

W@M

W@M Life Cycle Management
Improved productivity with information at your fingertips. Data relevant to aplant and its components is generated from the first stages of planning andduring the asset’s complete life cycle.
W@M Life Cycle Management is an open and flexible information platform
with online and on-site tools. Instant access for your staff to current, in-depth
data shortens your plant’s engineering time, speeds up procurement processes
and increases plant uptime.
Combined with the right services, W@M Life Cycle Management boosts
productivity in every phase. For more information, see:www.endress.com/lifecyclemanagement

FieldCare

FDT-based plant asset management tool from Endress+Hauser.
It can configure all smart field units in your system and helps you manage
them. By using the status information, it is also a simple but effective way ofchecking their status and condition.

 Operating Instructions BA00027S and BA00059S
DeviceCare

Tool for connecting and configuring Endress+Hauser field devices.

 Innovation brochure IN01047S

126

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

15.4

Accessories

System components

Accessories

Description

Memograph M graphic
data manager

The Memograph M graphic data manager provides information on all the relevantmeasured variables. Measured values are recorded correctly, limit values aremonitored and measuring points analyzed. The data are stored in the 256 MB
internal memory and also on a SD card or USB stick.
Information TI00133R
 •• Technical
Operating Instructions BA00247R

iTEMP

The temperature transmitters can be used in all applications and are suitable forthe measurement of gases, steam and liquids. They can be used to read in themedium temperature.

 "Fields of Activity" document FA00006T

Endress+Hauser

127

Technical data

Proline Promass P 100 PROFIBUS DP

16

Technical data

16.1

Application

The measuring device is intended only for the flow measurement of liquids.
Depending on the version ordered, the measuring device can also measure potentiallyexplosive, flammable, poisonous and oxidizing media.
To ensure that the device remains in proper operating condition for its service life, use themeasuring device only for media against which the process-wetted materials are
sufficiently resistant.

16.2

Function and system design

Measuring principle

Mass flow measurement based on the Coriolis measuring principle

Measuring system

The device consists of a transmitter and a sensor.
The device is available as a compact version:
The transmitter and sensor form a mechanical unit.
Information on the structure of the device →  11

128

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

16.3
Measured variable

Technical data

Input

Direct measured variables• Mass flow
• Density• Temperature
Calculated measured variables• Volume flow
• Corrected volume flow
• Reference density

Measuring range

Measuring range for liquids
DN

Measuring range full scale values min(F) to max(F)

[mm]

[in]

[kg/h][lb/min]

8³⁄₈

0 to 2 000

0 to 73.50

15½

0 to 6 500

0 to 238.9

25

1

0 to 18 000

0 to 661.5

40

1½

0 to 45 000

0 to 1 654

50

20 to 70 000

0 to 2 573

Recommended measuring range
Flow limit →  139

Operable flow range

Over 1000 : 1.
Flow rates above the preset full scale value do not override the electronics unit, with theresult that the totalizer values are registered correctly.

Input signal

External measured values
To increase the accuracy of certain measured variables or to calculate the corrected volume
flow for gases, the automation system can continuously write different measured values tothe measuring device:• Operating pressure to increase accuracy (Endress+Hauser recommends the use of apressure measuring device for absolute pressure, e.g. Cerabar M or Cerabar S)• Medium temperature to increase accuracy (e.g. iTEMP)• Reference density for calculating the corrected volume flow for gases
Various pressure transmitters and temperature measuring devices can be orderedfrom Endress+Hauser: see "Accessories" section →  127
It is recommended to read in external measured values to calculate the following measured
variables:• Mass flow
• Corrected volume flow
Digital communication
The measured values are written by the automation system via PROFIBUS DP.

Endress+Hauser

129Technical data

Proline Promass P 100 PROFIBUS DP

16.4
Output signal

Signal on alarm

OutputPROFIBUS DP
Signal encoding

NRZ code

Data transfer

9.6 kBaud…12 MBaud

Depending on the interface, failure information is displayed as follows.
PROFIBUS DP
Status and alarm
messages

Diagnostics in accordance with PROFIBUS PA Profile 3.02

Local display
Plain text display

With information on cause and remedial measures

Backlight

Red backlighting indicates a device error.

Status signal as per NAMUR recommendation NE 107
Interface/protocol• Via digital communication:
PROFIBUS DP• Via service interface
CDI-RJ45 service interface
Plain text display

With information on cause and remedial measures

Web browser
Plain text display

With information on cause and remedial measures

Light emitting diodes (LED)
Status information

Status indicated by various light emitting diodes
The following information is displayed depending on the device version:• Supply voltage active• Data transmission active• Device alarm/error has occurred Diagnostic information via light emitting diodes
Low flow cut off

The switch points for low flow cut off are user-selectable.

Galvanic isolation

The following connections are galvanically isolated from each other:• Outputs• Power supply

130

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Protocol-specific data

Technical data

protocol-specific data
Manufacturer ID

0x11

Ident number

0x1561

Profile version

3.02

Device description files (GSD,
DTM, DD)

Information and files at:• https://www.endress.com/download
On the device product page: PRODUCTS → Product Finder → Links• https://www.profibus.com

Output values(from measuring device toautomation system)

Analog input 1 to 8
• Mass flow
• Volume flow
• Corrected volume flow
• Target mass flow
• Carrier mass flow
• Density• Reference density• Concentration• Temperature• Carrier pipe temperature• Electronics temperature• Oscillation frequency• Oscillation amplitude• Frequency fluctuation• Oscillation damping• Tube damping fluctuation• Signal asymmetry• Exciter current
Digital input 1 to 2
• Partially filled pipe detection• Low flow cut off
Totalizer 1 to 3
• Mass flow
• Volume flow
• Corrected volume flow

Input values(from automation system tomeasuring device)

Analog output 1 to 3 (fixed assignment)• Pressure• Temperature• Reference density
Digital output 1 to 3 (fixed assignment)• Digital output 1: switch positive zero return on/off• Digital output 2: perform zero adjustment• Digital output 3: switch switch output on/off
Totalizer 1 to 3
• Totalize• Reset and hold
• Preset and hold
• Stop• Operating mode configuration:• Net flow total• Forward flow total• Reverse flow total

Endress+Hauser

Supported functions• Identification & Maintenance
Simplest device identification on the part of the control system andnameplate• PROFIBUS upload/download
Reading and writing parameters is up to ten times faster with PROFIBUS
upload/download• Condensed Status
Simplest and self-explanatory diagnostic information by categorizing
diagnostic messages that occur

Configuration of the device
address• DIP switches on the I/O electronics module• Via operating tools (e.g. FieldCare)131

Technical data

Proline Promass P 100 PROFIBUS DP

16.5

Power supply

Terminal assignment• →  27•

Supply voltage

The power unit must be tested to ensure it meets safety requirements (e.g. PELV, SELV).
Transmitter
DC 20 to 30 V

Power consumption

Current consumption

Transmitter
Order code for "Output"

Maximum
Power consumption

Option L: PROFIBUS DP

3.5 W

Transmitter
Order code for "Output"

Maximum
Current consumption

Maximumswitch-on current

Option L: PROFIBUS DP

145 mA

18 A (< 0.125 ms)

Device fuseFine-wire fuse (slow-blow) T2A

Power supply failure• Totalizers stop at the last value measured.
• Depending on the device version, the configuration is retained in the device memory orin the plug-in memory (HistoROM DAT).
• Error messages (incl. total operated hours) are stored.

Electrical connection→  29Potential equalization
Terminals

Transmitter
Spring terminals for wire cross-sections0.5 to 2.5 mm2 (20 to 14 AWG)

Cable entries• Cable gland: M20 × 1.5 with cable Ø 6 to 12 mm (0.24 to 0.47 in)• Thread for cable entry:• M20• G ½"• NPT ½"

Cable specification→  26132

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

16.6
Reference operating
conditions

Technical data

Performance characteristics• Error limits based on ISO 11631• Water with +15 to +45 °C (+59 to +113 °F) at 2 to 6 bar (29 to 87 psi)
• Specifications as per calibration protocol• Accuracy based on accredited calibration rigs that are traced to ISO 17025.
To obtain measured errors, use the Applicator sizing tool →  126

Maximum measured erroro.r. = of reading; 1 g/cm³ = 1 kg/l; T = medium temperature
Base accuracy
Design fundamentals →  136
Mass flow and volume flow (liquids)±0.10 % o.r.
Density (liquids)

1)
2)
3)

Under reference conditions

Standard density calibration 1)

Wide-range
Density specification 2) 3)[g/cm³][g/cm³][g/cm³]±0.0005±0.01±0.002

Valid over the entire temperature and density range
Valid range for special density calibration: 0 to 2 g/cm³, +10 to +80 °C (+50 to +176 °F)
order code for "Application package", option EE "Special density"

Temperature±0.5 °C ± 0.005 · T °C (±0.9 °F ± 0.003 · (T – 32) °F)
Zero point stability
DN

Zero point stability

[mm]

[in]

[kg/h][lb/min]

8

³⁄₈

0.20

0.007

15

½

0.65

0.024

25

1

1.80

0.066

40

1½

4.50

0.165

50

2

7.0

0.257

Flow values
Flow values as turndown parameters depending on the nominal diameter.

Endress+Hauser

133

Technical data

Proline Promass P 100 PROFIBUS DP

SI units
DN

1:1

1:10

1:20

1:50

1:100

1:500

[mm]

[kg/h][kg/h][kg/h][kg/h][kg/h][kg/h]

8

2 000

200

100

40

20

4

15

6 500

650

325

130

65

13

25

18 000

1 800

900

360

180

36

40

45 000

4 500

2 250

900

450

90

50

70 000

7 000

3 500

1 400

700

140

DN

1:1

1:10

1:20

1:50

1:100

1:500

[inch][lb/min][lb/min][lb/min][lb/min][lb/min][lb/min]

³⁄₈

73.50

7.350

3.675

1.470

0.735

0.147

½

238.9

23.89

11.95

4.778

2.389

0.478

1

661.5

66.15

33.08

13.23

6.615

1.323

1½

1 654

165.4

82.70

33.08

16.54

3.308

2

2 573

257.3

128.7

51.46

25.73

5.146

US units

Accuracy of outputs
The output accuracy must be factored into the measured error if analog outputs areused, but can be ignored for fieldbus outputs (e.g. Modbus RS485, EtherNet/IP).
The outputs have the following base accuracy specifications.
Repeatabilityo.r. = of reading; 1 g/cm3 = 1 kg/l; T = medium temperature
Base repeatability
Design fundamentals →  136
Mass flow and volume flow (liquids)±0.05 % o.r.
Density (liquids)±0.00025 g/cm3
Temperature±0.25 °C ± 0.0025 · T °C (±0.45 °F ± 0.0015 · (T–32) °F)

Response time

The response time depends on the configuration (damping).

Influence of medium
temperature

Mass flow and volume flowo.f.s. = of full scale value
If there is a difference between the temperature during zero adjustment and the processtemperature, the additional measured error of the sensors is typically ±0.0002 %o.f.s./°C(±0.0001 % o.f.s./°F).

134

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Technical data

The influence is reduced when the zero adjustment is performed at process temperature.
Density
If there is a difference between the density calibration temperature and the processtemperature, the measured error of the sensors is typically±0.0001 g/cm3/°C (±0.00005 g/cm3/°F). Field density adjustment is possible.
Wide-range density specification (special density calibration)
If the process temperature is outside the valid range (→  133) the measured error is±0.0001 g/cm3 /°C (±0.00005 g/cm3 /°F)
3[kg/m ]
20
18
16
14
12
10

1

8
6

2

4
2
0
-50
-80 -40

0
0

50

100

150

200 [°C]

40 80 120 160 200 240 280 320 360 400 [°F]
A0016610

12

Field density adjustment, for example at +20 °C (+68 °F)
Special density calibration

Temperature±0.005 · T °C (± 0.005 · (T – 32) °F)
Influence of medium
pressure

The tables below show the effect that a difference in pressure between the calibrationpressure and the process pressure has on the accuracy in the case of the mass flow anddensity.
o.r. = of reading
It is possible to compensate for the effect by:• Reading in the current pressure measured value via the current input or a digital
input.
• Specifying a fixed value for the pressure in the device parameters.
Operating Instructions .
DN

Endress+Hauser[% o.r./bar][% o.r./psi]

[mm]

[in]

8

³⁄₈

–0.002–0.0001

15½–0.006–0.000425

1–0.005–0.000340

1½–0.007–0.000550

2–0.006–0.0004135

Technical data

Design fundamentals

Proline Promass P 100 PROFIBUS DP

o.r. = of reading, o.f.s. = of full scale value
BaseAccu = base accuracy in % o.r., BaseRepeat = base repeatability in % o.r.
MeasValue = measured value; ZeroPoint = zero point stability
Calculation of the maximum measured error as a function of the flow rate
Flow rate

Maximum measured error in % o.r.

ZeroPoint³ BaseAccu ⋅ 100± BaseAccu
A0021339
A0021332

ZeroPoint< BaseAccu ⋅ 100±

ZeroPoint
MeasValue ⋅ 100

A0021333

A0021334

Calculation of the maximum repeatability as a function of the flow rate
Flow rate³

Maximum repeatability in % o.r.

½ ⋅ ZeroPoint
BaseRepeat ⋅ 100

A0021340
A0021335<½ ⋅ ZeroPoint
BaseRepeat ⋅ 100

ZeroPoint± ½ ⋅ MeasValue ⋅ 100
A0021337

A0021336

Example of maximum measured error
E [%]
2.5
2.0
1.5
1.0
0.5
0
0

10

20

30

40

50

60

70

80

90

100 Q [%]
A0030317

E
Q

Maximum measured error in % o.r. (example)
Flow rate in % of maximum full scale value

16.7
Mounting requirements→  18

16.8
Ambient temperature
range

Mounting

Environment→  20→  20
Temperature tables
Observe the interdependencies between the permitted ambient and fluid
temperatures when operating the device in hazardous areas.
For detailed information on the temperature tables, see the separate document
entitled "Safety Instructions" (XA) for the device.

136

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Storage temperature

Technical data

–40 to +80 °C (–40 to +176 °F), preferably at +20 °C (+68 °F) (standard version)–50 to +80 °C (–58 to +176 °F) (Order code for "Test, certificate", option JM)

Climate class

DIN EN 60068-2-38 (test Z/AD)

Degree of protection

Transmitter and sensor• Standard: IP66/67, Type 4X enclosure, suitable for pollution degree 4• With the order code for "Sensor options", option CM: IP69 can also be ordered• When the housing is open: IP20, Type 1 enclosure, suitable for pollution degree 2• Display module: IP20, Type 1 enclosure, suitable for pollution degree 2

Vibration- and shockresistanceVibration sinusoidal, in accordance with IEC 60068-2-6• 2 to 8.4 Hz, 3.5 mm peak
• 8.4 to 2 000 Hz, 1 g peak
Vibration broad-band random, according to IEC 60068-2-64• 10 to 200 Hz, 0.003 g2/Hz• 200 to 2 000 Hz, 0.001 g2/Hz• Total: 1.54 g rms
Shock half-sine, according to IEC 60068-2-27
6 ms 30 g
Rough handling shocks, according to IEC 60068-2-31

Interior cleaning• Cleaning in place (CIP)• Sterilization in place (SIP)• Cleaning with pigs
Options
Oil- and grease-free version for wetted parts, without declaration
Order code for "Service", option HA

Electromagnetic
compatibility (EMC)• As per IEC/EN 61326 and NAMUR Recommendation 21 (NE 21)• Complies with emission limits for industry as per EN 55011 (Class A)• Device version with PROFIBUS DP: Complies with emission limits for industry as per EN
50170 Volume 2, IEC 61784
The following applies for PROFIBUS DP: If baud rates > 1.5 MBaud, an EMC cableentry must be used and the cable shield must continue as far as the terminal wherever
possible.
Details are provided in the Declaration of Conformity.
This unit is not intended for use in residential environments and cannot guarantee
adequate protection of the radio reception in such environments.

16.9

Process

Medium temperature range

Endress+Hauser

137

Technical data

Proline Promass P 100 PROFIBUS DP

Standard version–50 to +150 °C (–58 to +302 °F)

Order code for "Measuring tube
mat., wetted surface", option BB,
BC, BD

Extended temperature version–50 to +205 °C (–58 to +401 °F)

Order code for "Measuring tube
mat., wetted surface", option TD, TG

Dependency of ambient temperature on medium temperature
Ta
A
B

Tm
A0031121

 17
Ta
Tm
A
B

Exemplary representation, values in the table below.

Ambient temperature
Medium temperature
Maximum permitted medium temperature Tm at Ta max = 60 °C (140 °F); higher medium temperatures Tm
require a reduction in the ambient temperature Ta
Maximum permitted ambient temperature Ta for the maximum specified medium temperature Tm of thesensorValues for devices that are used in the hazardous area:
Separate Ex documentation (XA) for the device .
Density
Pressure-temperature
ratings
Sensor housing

0 to 5 000 kg/m3 (0 to 312 lb/cf)
For an overview of the pressure-temperature ratings for the process connections, see
the Technical Information

The sensor housing is filled with dry nitrogen gas and protects the electronics andmechanics inside.
If a measuring tube fails (e.g. due to process characteristics like corrosive or abrasive
fluids), the fluid will initially be contained by the sensor housing.
If the sensor is to be purged with gas (gas detection), it should be equipped with purge
connections.
Do not open the purge connections unless the containment can be filled immediately
with a dry, inert gas. Use only low pressure to purge.
Maximum pressure: 5 bar (72.5 psi)

138

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Technical data

Burst pressure of the sensor housing
The following sensor housing burst pressures are only valid for standard devices and/or
devices equipped with closed purge connections (not opened/as delivered).
If a device fitted with purge connections (order code for "Sensor option", option CH "Purge
connection") is connected to the purge system, the maximum pressure is determined by thepurge system itself or by the device, depending on which component has the lower
pressure classification.
The sensor housing burst pressure refers to a typical internal pressure which is reachedprior to mechanical failure of the sensor housing and which was determined during type
testing. The corresponding type test declaration can be ordered with the device (order codefor "Additional approval", option LN "Sensor housing burst pressure, type test").
DN

Sensor housing burst pressure

[mm]

[in]

[bar]

[psi]

8

³⁄₈

190

2 755

15

½

175

2 538

25

1

165

2 392

40

1½

152

2 204

50

2

103

1 494

For information on the dimensions: see the "Mechanical construction" section of the"Technical Information" document
Flow limit

Select the nominal diameter by optimizing between the required flow range andpermissible pressure loss.
For an overview of the full scale values for the measuring range, see the "Measuring
range" section →  129• The minimum recommended full scale value is approx. 1/20 of the maximum full scale
value• In most applications, 20 to 50 % of the maximum full scale value can be considered ideal• A low full scale value must be selected for abrasive media (such as liquids with entrainedsolids): flow velocity < 1 m/s (< 3 ft/s).
To calculate the flow limit, use the Applicator sizing tool →  126

Pressure loss

System pressure

Endress+Hauser

To calculate the pressure loss, use the Applicator sizing tool →  126→  20139

Technical data

Proline Promass P 100 PROFIBUS DP

16.10 Mechanical construction
Design, dimensions

Weight

For the dimensions and installation lengths of the device, see the "Technical
Information" document, "Mechanical construction" section .
All values (weight exclusive of packaging material) refer to devices with EN/DIN
PN 40 flanges. Weight specifications including transmitter: order code for "Housing",
option A "Compact, aluminum coated".
Weight in SI units
DN[mm]

Weight [kg]

8

12

15

14

25

20

40

36

50

59

Weight in US units

Materials

DN[in]

Weight [lbs]

3/8

26

½

31

1

44

1½

79

2

130

Transmitter housing• Order code for "Housing", option A "Compact, aluminum coated":
Aluminum, AlSi10Mg, coated• Order code for "Housing", option B "Compact, hygienic, stainless":
Hygienic version, stainless steel 1.4301 (304)• Order code for "Housing", option C "Ultra-compact, hygienic, stainless":
Hygienic version, stainless steel 1.4301 (304)• Window material for optional local display (→  143):• For order code for "Housing", option A: glass• For order code for "Housing", option B and C: plastic

140

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Technical data

Cable entries/cable glands

1

2
3
A0020640

 18
1
2
3

Possible cable entries/cable glands

Female thread M20 × 1.5
Cable gland M20 × 1.5
Adapter for cable entry with female thread G ½" or NPT ½"

Order code for "Housing", option A "Compact, aluminum, coated"
The various cable entries are suitable for hazardous and non-hazardous areas.
Cable entry/cable glandMaterial

Cable gland M20 × 1.5
Adapter for cable entry with internal thread G ½"

Nickel-plated brass

Adapter for cable entry with internal thread NPT ½"

Order code for "Housing", option B "Compact, hygienic, stainless"
The various cable entries are suitable for hazardous and non-hazardous areas.
Cable entry/cable glandMaterial

Cable gland M20 × 1.5

Stainless steel, 1.4404 (316L)

Adapter for cable entry with internal thread G ½"
Adapter for cable entry with internal thread NPT ½"

Device plug
Electrical connection
Plug M12x1

Material• Socket: Stainless steel, 1.4404 (316L)• Contact housing: Polyamide• Contacts: Gold-plated brass

Sensor housing• Acid and alkali-resistant outer surface• Stainless steel 1.4301 (304)
Measuring tubes
Stainless steel, 1.4435 BN2 (316L)

Endress+Hauser

141

Technical data

Proline Promass P 100 PROFIBUS DP

Process connections• Flanges according to EN 1092-1 (DIN 2501) / according to ASME B16.5 / according to
JIS B2220:
Stainless steel, 1.4404 (F316/F316L)• All other process connections:
Stainless steel, 1.4435 BN2 (316L)
Available process connections→  142
Seals
Welded process connections without internal seals
Accessories
Protective cover
Stainless steel, 1.4404 (316L)
Safety Barrier Promass 100
Housing: Polyamide
Process connections• Fixed flange connections:• EN 1092-1 (DIN 2501) flange• EN 1092-1 (DIN 2512N) flange• ASME B16.5 flange• JIS B2220 flange• DIN 11864-2 Form A flange, DIN 11866 series A, flange with notch• BBS flange small (sterile orbital), DIN 11866 series A, female• BBS flange small (sterile orbital), DIN 11866 series B, female• Clamp connections:• Tri-Clamp (OD tubes), DIN 11866 series C• DIN 11864-3 Form A clamp, DIN 11866 series A, with notch• DIN 32676 clamp, DIN 11866 series A• ISO 2852 clamp, ISO 2037
• ISO 2852 clamp, DIN 11866 series B• BBS Quick-Connect (sterile orbital), DIN 11866 series A, female• BBS Quick-Connect (sterile orbital), DIN 11866 series B, female• Neumo BioConnect clamp, DIN 11866 series A, clamp form R• Eccentric clamp connection:• Eccen. Tri-Clamp, DIN 11866 series C• DIN 11864-3 Form A clamp, DIN 11866 series A, with notch• DIN 32676 clamp, DIN 11866 series A• ISO 2852 clamp, DIN 11866 series B• BBS Quick-Connect (sterile orbital), DIN 11866 series A, female• BBS Quick-Connect (sterile orbital), DIN 11866 series B, female• Neumo BioConnect clamp, DIN 11866 series A, clamp form R• Thread:• DIN 11851 thread, DIN 11866 series A• SMS 1145 thread• ISO 2853 thread, ISO 2037
• DIN 11864-1 Form A thread, DIN 11866 series A• BBS thread (sterile orbital), DIN 11866 series A• BBS thread (sterile orbital), DIN 11866 series B
Process connection materials

142

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Surface roughness

Technical data

All data refer to parts in contact with the medium. The following surface roughness
categories can be ordered.
• Ramax = 0.76 µm (30 µin)
• Ramax = 0.38 µm (15 µin) electropolished• Delta ferrite <1%

16.11 Operability
Local display

The local display is only available with the following device order code:
Order code for "Display; operation", option B: 4-line; illuminated, via communication
Display element• 4-line liquid crystal display with 16 characters per line.
• White background lighting; switches to red in event of device errors.
• Format for displaying measured variables and status variables can be individuallyconfigured.
• Permitted ambient temperature for the display: –20 to +60 °C (–4 to +140 °F). The
readability of the display may be impaired at temperatures outside the temperature
range.
Disconnecting the local display from the main electronics module
In the case of the "Compact, aluminum coated" housing version, the local display must
only be disconnected manually from the main electronics module. In the case of the"Compact, hygienic, stainless" and "Ultra-compact, hygienic, stainless" housing
versions, the local display is integrated in the housing cover and is disconnected fromthe main electronics module when the housing cover is opened.
"Compact, aluminum coated" housing version
The local display is plugged onto the main electronics module. The electronic connection
between the local display and main electronics module is established via a connecting
cable.
For some work performed on the measuring device (e.g. electrical connection), it isadvisable to disconnect the local display from the main electronics module:
1.

Press in the side latches of the local display.

2.

Remove the local display from the main electronics module. Pay attention to thelength of the connecting cable when doing so.

Once the work is completed, plug the local display back on.
Remote operation

Via PROFIBUS DP network
This communication interface is available in device versions with PROFIBUS DP.

Endress+Hauser

143

Technical data

Proline Promass P 100 PROFIBUS DP

1

2

3

4

4

4
A0020903

 19
1
2
3
4

Service interface

Options for remote operation via PROFIBUS DP network

Automation system
Computer with PROFIBUS network card
PROFIBUS DP network
Measuring device

Via service interface (CDI-RJ45)
PROFIBUS DP23

1
A0021270

 20
1
2
3

Languages

144

Connection for order code for "Output", option L: PROFIBUS DP

Service interface (CDI-RJ45) of the measuring device with access to the integrated web server
Computer with web browser (e.g. Internet Explorer) for accessing the integrated web server or with "FieldCare"
operating tool with COM DTM "CDI Communication TCP/IP"
Standard Ethernet connecting cable with RJ45 plug

Can be operated in the following languages:• Via "FieldCare" operating tool: English, German, French, Spanish, Italian, Chinese,
Japanese• Via Web browser
English, German, French, Spanish, Italian, Dutch, Portuguese, Polish, Russian, Turkish,
Chinese, Japanese, Bahasa (Indonesian), Vietnamese, Czech, Swedish, KoreanEndress+Hauser

Proline Promass P 100 PROFIBUS DP

Technical data

16.12 Certificates and approvals
Current certificates and approvals that are available for the product can be selected via the
Product Configurator at www.endress.com:

CE mark

1.

Select the product using the filters and search field.

2.

Open the product page.

3.

Select Configuration.

The device meets the legal requirements of the applicable EU Directives. These are listed in
the corresponding EU Declaration of Conformity along with the standards applied.
Endress+Hauser confirms successful testing of the device by affixing to it the CE mark.

UKCA marking

The device meets the legal requirements of the applicable UK regulations (Statutory
Instruments). These are listed in the UKCA Declaration of Conformity along with thedesignated standards. By selecting the order option for UKCA marking, Endress+Hauser
confirms a successful evaluation and testing of the device by affixing the UKCA mark.
Contact address Endress+Hauser UK:
Endress+Hauser Ltd.
Floats Road
Manchester M23 9NF
United Kingdom
www.uk.endress.com

RCM mark

The measuring system meets the EMC requirements of the "Australian Communications
and Media Authority (ACMA)".

Ex approval

The devices are certified for use in hazardous areas and the relevant safety instructions areprovided in the separate "Safety Instructions" (XA) document. Reference is made to thisdocument on the nameplate.

Hygienic compatibility• 3-A approval• Only measuring devices with the order code for "Additional approval", option LP "3A"
have 3-A approval.
• The 3-A approval refers to the measuring device.
• When installing the measuring device, ensure that no liquid can accumulate on theoutside of the measuring device.
A remote display module must be installed in accordance with the 3-A Standard.
• Accessories (e.g. heating jacket, weather protection cover, wall holder unit) must be
installed in accordance with the 3-A Standard.
Each accessory can be cleaned. Disassembly may be necessary under certain
circumstances.
• EHEDG-tested
Only devices with the order code for "Additional approval", option LT "EHEDG" have beentested and meet the requirements of the EHEDG.
To meet the requirements for EHEDG certification, the device must be used with processconnections in accordance with the EHEDG position paper entitled "Easy cleanable Pipe
couplings and Process connections" (www.ehedg.org).

Pharmaceuticalcompatibility• FDA 21 CFR 177• USP <87>
• USP <88> Class VI 121 °C• TSE/BSE Certificate of Suitability

Endress+Hauser

145

Technical data

Certification PROFIBUS

Proline Promass P 100 PROFIBUS DPPROFIBUS interface
The measuring device is certified and registered by the PNO (PROFIBUS
Nutzerorganisation e.V./PROFIBUS User Organization). The measuring system meets all
the requirements of the following specifications:• Certified according to PA Profile 3.02
• The device can also be operated with certified devices of other manufacturers(interoperability)

Pressure Equipment
Directive• With the marking:a) PED/G1/x (x = category) orb) UK/G1/x (x = category)on the sensor nameplate, Endress+Hauser confirms compliance with the "Essential
Safety Requirements"a) specified in Annex I of the Pressure Equipment Directive 2014/68/EU orb) Schedule 2 of Statutory Instruments 2016 No. 1105.
• Devices not bearing this marking (without PED or UKCA) are designed andmanufactured according to sound engineering practice. They meet the requirements ofa) Art. 4 Para. 3 of the Pressure Equipment Directive 2014/68/EU orb) Part 1, Para. 8 of Statutory Instruments 2016 No. 1105.
The scope of application is indicateda) in diagrams 6 to 9 in Annex II of the Pressure Equipment Directive 2014/68/EU orb) Schedule 3, Para. 2 of Statutory Instruments 2016 No. 1105.

Other standards andguidelines• EN 60529
Degrees of protection provided by enclosures (IP code)• IEC/EN 60068-2-6
Environmental influences: Test procedure - Test Fc: vibrate (sinusoidal).
• IEC/EN 60068-2-31
Environmental influences: Test procedure - Test Ec: shocks due to rough handling,
primarily for devices.
• EN 61010-1
Safety requirements for electrical equipment for measurement, control and laboratoryuse - general requirements• IEC/EN 61326-2-3
Emission in accordance with Class A requirements. Electromagnetic compatibility (EMC
requirements).
• NAMUR NE 21
Electromagnetic compatibility (EMC) of industrial process and laboratory control
equipment• NAMUR NE 32
Data retention in the event of a power failure in field and control instruments withmicroprocessors• NAMUR NE 43
Standardization of the signal level for the breakdown information of digital transmitterswith analog output signal.
• NAMUR NE 53
Software of field devices and signal-processing devices with digital electronics• NAMUR NE 80
The application of the pressure equipment directive to process control devices• NAMUR NE 105
Specifications for integrating fieldbus devices in engineering tools for field devices• NAMUR NE 107
Self-monitoring and diagnosis of field devices• NAMUR NE 131
Requirements for field devices for standard applications

146

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Technical data

• NAMUR NE 132
Coriolis mass meter• ETSI EN 300 328
Guidelines for 2.4 GHz radio components.
• EN 301489
Electromagnetic compatibility and radio spectrum matters (ERM).

16.13 Application packages
Many different application packages are available to enhance the functionality of thedevice. Such packages might be needed to address safety aspects or specific application
requirements.
The application packages can be ordered with the device or subsequently fromEndress+Hauser. Detailed information on the order code in question is available from yourlocal Endress+Hauser sales center or on the product page of the Endress+Hauser website:www.endress.com.
Detailed information on the application packages:
Special Documentation for the device →  148
Heartbeat Technology

Order code for "Application package", option EB "Heartbeat Verification + Monitoring"
Heartbeat Verification
Meets the requirement for traceable verification to DIN ISO 9001:2008 Chapter 7.6 a)"Control of monitoring and measuring equipment".
• Functional testing in the installed state without interrupting the process.
• Traceable verification results on request, including a report.
• Simple testing process via local operation or other operating interfaces.
• Clear measuring point assessment (pass/fail) with high test coverage within theframework of manufacturer specifications.
• Extension of calibration intervals according to operator's risk assessment.
Heartbeat Monitoring
Continuously supplies data, which are characteristic of the measuring principle, to an
external condition monitoring system for the purpose of preventive maintenance orprocess analysis. These data enable the operator to:• Draw conclusions - using these data and other information - about the impact process
influences (e.g. corrosion, abrasion, formation of buildup etc.) have on the measuringperformance over time.
• Schedule servicing in time.
• Monitor the process or product quality, e.g. gas pockets.
For detailed information, see the Special Documentation for the device.

Concentration
measurement

Order code for "Application package", option ED "Concentration"
Calculation and outputting of fluid concentrations.
The measured density is converted to the concentration of a substance of a binary mixture
using the "Concentration" application package:• Choice of predefined fluids (e.g. various sugar solutions, acids, alkalis, salts, ethanoletc.).
• Common or user-defined units (°Brix, °Plato, % mass, % volume, mol/l etc.) for standard
applications.
• Concentration calculation from user-defined tables.
The measured values are output via the digital and analog outputs of the device.
For detailed information, see the Special Documentation for the device.

Endress+Hauser

147

Technical data

Special density

Proline Promass P 100 PROFIBUS DP

Order code for "Application package", option EE "Special density"
Many applications use density as a key measured value for monitoring quality orcontrolling processes. The device measures the density of the fluid as standard and makes
this value available to the control system.
The "Special Density" application package offers high-precision density measurement over awide density and temperature range particularly for applications subject to varying process
conditions.
For detailed information, see the Operating Instructions for the device.

16.14 Accessories
Overview of accessories available for order →  125

16.15 Supplementary documentation
For an overview of the scope of the associated Technical Documentation, refer to thefollowing:• Device Viewer (www.endress.com/deviceviewer): Enter the serial number from thenameplate• Endress+Hauser Operations app: Enter serial number from nameplate or scanmatrix code on nameplate.
Standard documentation

Brief Operating Instructions

Brief Operating Instructions for the sensor
Measuring device

Documentation code

Proline Promass P

KA01286D

Transmitter Brief Operating Instructions
Measuring device

Documentation code

Proline Promass 100

KA01333D

Technical Information
Measuring device

Documentation code

Proline Promass P 100

TI01036D

Description of Device Parameters

Supplementary devicedependent documentation

148

Measuring device

Documentation code

Proline Promass 100

GP01034D

Safety Instructions
Content

Documentation code

ATEX/IECEx Ex i

XA00159D

ATEX/IECEx Ex nA

XA01029D

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Technical data

Content

Documentation code

cCSAus IS

XA00160D

INMETRO Ex i

XA01219D

INMETRO Ex nA

XA01220D

NEPSI Ex i

XA01249D

NEPSI Ex nA

XA01262D

Special Documentation
Content

Documentation code

Information on the Pressure Equipment Directive

SD00142D

Concentration measurement

SD01152D

Heartbeat Technology

SD01153D

Web server

SD01821D

Installation Instructions
Contents

Comment

Installation instructions for spare part sets andaccessories• Access the overview of all the available spare part sets via Device Viewer →  123• Accessories available for order with Installation Instructions →  125

Endress+Hauser

149

Index

Proline Promass P 100 PROFIBUS DP

Index
0…9

3-A approval . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 145

A

Access authorization to parameters
Read access . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 39
Write access . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 39
Access code . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 39
Incorrect input . . . . . . . . . . . . . . . . . . . . . . . . . . . 39
Adapting the diagnostic behavior . . . . . . . . . . . . . . . . 87
Analog Input module . . . . . . . . . . . . . . . . . . . . . . . . . 53
Analog Output module . . . . . . . . . . . . . . . . . . . . . . . 55
Application . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 128
Application packages . . . . . . . . . . . . . . . . . . . . . . . . 147
Approvals . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 145

C

Cable entries
Technical data . . . . . . . . . . . . . . . . . . . . . . . . . . 132
Cable entry
Degree of protection . . . . . . . . . . . . . . . . . . . . . . . 33
CE mark . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 10, 145
Certificates . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 145
Certification PROFIBUS . . . . . . . . . . . . . . . . . . . . . . 146
cGMP . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 145
Check
Connection . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 33
Installation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 25
Checklist
Post-connection check . . . . . . . . . . . . . . . . . . . . . 33
Post-installation check . . . . . . . . . . . . . . . . . . . . . 25
Cleaning
Cleaning in place (CIP) . . . . . . . . . . . . . . . . . . . . 122
Exterior cleaning . . . . . . . . . . . . . . . . . . . . . . . . 122
Interior cleaning . . . . . . . . . . . . . . . . . . . . . . . . . 122
Sterilization in place (SIP) . . . . . . . . . . . . . . . . . . 122
Cleaning in place (CIP) . . . . . . . . . . . . . . . . . . . . . . . 137
Climate class . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 137
Commissioning . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 58
Advanced settings . . . . . . . . . . . . . . . . . . . . . . . . 67
Configuring the measuring device . . . . . . . . . . . . . 58
Compatibility with previous model . . . . . . . . . . . . . . . 48
Connecting cable . . . . . . . . . . . . . . . . . . . . . . . . . . . . 26
Connecting the measuring device . . . . . . . . . . . . . . . . 29
Connection
see Electrical connection
Connection examples, potential equalization . . . . . . . 31
Connection preparations . . . . . . . . . . . . . . . . . . . . . . 29
Connection tools . . . . . . . . . . . . . . . . . . . . . . . . . . . . 26
Current consumption . . . . . . . . . . . . . . . . . . . . . . . . 132
Cyclic data transmission . . . . . . . . . . . . . . . . . . . . . . 52

D

Date of manufacture . . . . . . . . . . . . . . . . . . . . . . 13, 14
Deactivating the write protection . . . . . . . . . . . . . . . . 76
Declaration of Conformity . . . . . . . . . . . . . . . . . . . . . 10
150

Define access code . . . . . . . . . . . . . . . . . . . . . . . . . . . 76
Degree of protection . . . . . . . . . . . . . . . . . . . . . 33, 137
Density . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 138
Density adjustment . . . . . . . . . . . . . . . . . . . . . . . . . . 69
DesignMeasuring device . . . . . . . . . . . . . . . . . . . . . . . . . 11
Design fundamentals
Maximum measured error . . . . . . . . . . . . . . . . . 136
Repeatability . . . . . . . . . . . . . . . . . . . . . . . . . . . 136
Device components . . . . . . . . . . . . . . . . . . . . . . . . . . 11
Device description files . . . . . . . . . . . . . . . . . . . . . . . 48
Device fuse . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 132
Device locking, status . . . . . . . . . . . . . . . . . . . . . . . . 78
Device master file
GSD . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 48
Device nameSensor . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 14
Transmitter . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 13
Device repair . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 123
Device revision . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 48
Device type ID . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 48
Device Viewer . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 123
DeviceCare . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 47
Device description file . . . . . . . . . . . . . . . . . . . . . . 48
Diagnostic information
Design, description . . . . . . . . . . . . . . . . . . . . . 85, 87
DeviceCare . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 86
FieldCare . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 86
Light emitting diodes . . . . . . . . . . . . . . . . . . . . . . 84
Overview . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 90
Remedial measures . . . . . . . . . . . . . . . . . . . . . . . 90
Web browser . . . . . . . . . . . . . . . . . . . . . . . . . . . . 85
Diagnostic list . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 116
Dimensions . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 20
DIP switch
see Write protection switch
Discrete Input module . . . . . . . . . . . . . . . . . . . . . . . . 56
Discrete Output module . . . . . . . . . . . . . . . . . . . . . . . 57
Display area
For operational display . . . . . . . . . . . . . . . . . . . . . 38
Display values
For locking status . . . . . . . . . . . . . . . . . . . . . . . . . 78
Disposal . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 124
Document
Function . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6
Symbols . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6
Document function . . . . . . . . . . . . . . . . . . . . . . . . . . . 6
Document information . . . . . . . . . . . . . . . . . . . . . . . . 6
Down pipe . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 18

E

EHDEG-certified . . . . . . . . . . . . . . . . . . . . . . . . . . . 145
Electrical connection
Degree of protection . . . . . . . . . . . . . . . . . . . . . . . 33
Measuring device . . . . . . . . . . . . . . . . . . . . . . . . . 26

Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Operating tools
Via PROFIBUS DP network . . . . . . . . . . . . 45, 143
Via service interface (CDI-RJ45) . . . . . . . . 45, 144
Web server . . . . . . . . . . . . . . . . . . . . . . . . . . 45, 144
Electromagnetic compatibility . . . . . . . . . . . . . . . . . 137
EMPTY_MODULE module . . . . . . . . . . . . . . . . . . . . . 57
Enabling write protection . . . . . . . . . . . . . . . . . . . . . 76
Endress+Hauser services
Maintenance . . . . . . . . . . . . . . . . . . . . . . . . . . . 122
Repair . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 123
Environment
Storage temperature . . . . . . . . . . . . . . . . . . . . . . 137
Vibration- and shock-resistance . . . . . . . . . . . . . 137
Error messages
see Diagnostic messages
Event list . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 116
Event logbook . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 116
Ex approval . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 145
Extended order code
Sensor . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 14
Transmitter . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 13Exterior cleaning . . . . . . . . . . . . . . . . . . . . . . . . . . . 122

F

FDA . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 145
Field of application
Residual risks . . . . . . . . . . . . . . . . . . . . . . . . . . . . 10
FieldCare . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 46
Device description file . . . . . . . . . . . . . . . . . . . . . . 48
Establishing a connection . . . . . . . . . . . . . . . . . . . 46
Function . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 46
User interface . . . . . . . . . . . . . . . . . . . . . . . . . . . 47
Filtering the event logbook . . . . . . . . . . . . . . . . . . . 117
FirmwareRelease date . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 48
Version . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 48
Firmware history . . . . . . . . . . . . . . . . . . . . . . . . . . . 121
Flow direction . . . . . . . . . . . . . . . . . . . . . . . . . . . 19, 23
Flow limit . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 139
Food Contact Materials Regulation . . . . . . . . . . . . . . 145
FOUNDATION Fieldbus block structure . . . . . . . . . . . 50
Functionssee Parameters

G

Galvanic isolation . . . . . . . . . . . . . . . . . . . . . . . . . . 130

H

Hardware write protection . . . . . . . . . . . . . . . . . . . . . 77
Hygienic compatibility . . . . . . . . . . . . . . . . . . . . . . . 145

I

I/O electronics module . . . . . . . . . . . . . . . . . . . . . 11, 29
Identifying the measuring device . . . . . . . . . . . . . . . . 12
Incoming acceptance . . . . . . . . . . . . . . . . . . . . . . . . . 12
Influence
Medium pressure . . . . . . . . . . . . . . . . . . . . . . . . 135
Medium temperature . . . . . . . . . . . . . . . . . . . . . 134
Inlet runs . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 20
Input . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 129
Endress+Hauser

Index

Inspection
Received goods . . . . . . . . . . . . . . . . . . . . . . . . . . . 12
Intended use . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9Interior cleaning . . . . . . . . . . . . . . . . . . . . . . . 122, 137

L

Languages, operation options . . . . . . . . . . . . . . . . . 144
Local display
see Operational display
Low flow cut off . . . . . . . . . . . . . . . . . . . . . . . . . . . 130

M

Main electronics module . . . . . . . . . . . . . . . . . . . . . . 11
Maintenance tasks . . . . . . . . . . . . . . . . . . . . . . . . . 122
Manufacturer ID . . . . . . . . . . . . . . . . . . . . . . . . . . . . 48
Materials . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 140
Maximum measured error . . . . . . . . . . . . . . . . . . . . 133
Measured values
see Process variables
Measurement accuracy . . . . . . . . . . . . . . . . . . . . . . 133
Measuring and test equipment . . . . . . . . . . . . . . . . 122
Measuring device
Configuration . . . . . . . . . . . . . . . . . . . . . . . . . . . . 58
Conversion . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 123
Design . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 11
Disposal . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 124
Mounting the sensor . . . . . . . . . . . . . . . . . . . . . . 23
Preparing for electrical connection . . . . . . . . . . . . 29
Preparing for mounting . . . . . . . . . . . . . . . . . . . . 23
Removing . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 124
Repairs . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 123
Measuring principle . . . . . . . . . . . . . . . . . . . . . . . . 128
Measuring range
For liquids . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 129
Measuring range, recommended . . . . . . . . . . . . . . . 139
Measuring system . . . . . . . . . . . . . . . . . . . . . . . . . . 128
Medium pressure
Influence . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 135
Medium temperature
Influence . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 134
Menu
Diagnostics . . . . . . . . . . . . . . . . . . . . . . . . . . . . 115
Operation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 78
Setup . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 59
Menus
For measuring device configuration . . . . . . . . . . . . 58
For specific settings . . . . . . . . . . . . . . . . . . . . . . . 67
ModuleAnalog Input . . . . . . . . . . . . . . . . . . . . . . . . . . . . 53
Analog output . . . . . . . . . . . . . . . . . . . . . . . . . . . 55
Discrete Input . . . . . . . . . . . . . . . . . . . . . . . . . . . . 56
Discrete Output . . . . . . . . . . . . . . . . . . . . . . . . . . 57
EMPTY_MODULE . . . . . . . . . . . . . . . . . . . . . . . . 57
Totalizer
SETTOT_MODETOT_TOTAL . . . . . . . . . . . . . . 55
SETTOT_TOTAL . . . . . . . . . . . . . . . . . . . . . . . 54
TOTAL . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 54
Mounting . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 18

151

IndexMounting dimensions
see Dimensions
Mounting location . . . . . . . . . . . . . . . . . . . . . . . . . . . 18
Mounting preparations . . . . . . . . . . . . . . . . . . . . . . . 23
Mounting requirements
Dimensions . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 20
Down pipe . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 18
Inlet and outlet runs . . . . . . . . . . . . . . . . . . . . . . . 20
Mounting location . . . . . . . . . . . . . . . . . . . . . . . . 18
Orientation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 19
Sensor heating . . . . . . . . . . . . . . . . . . . . . . . . . . . 21
System pressure . . . . . . . . . . . . . . . . . . . . . . . . . . 20
Thermal insulation . . . . . . . . . . . . . . . . . . . . . . . . 20
Vibrations . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 22
Mounting tool . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 23

N

NameplateSensor . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 14
Transmitter . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 13

O

Operable flow range . . . . . . . . . . . . . . . . . . . . . . . . 129
Operating menu
Menus, submenus . . . . . . . . . . . . . . . . . . . . . . . . 36
Structure . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 36
Submenus and user roles . . . . . . . . . . . . . . . . . . . 37
Operating philosophy . . . . . . . . . . . . . . . . . . . . . . . . 37
Operation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 78
Operation options . . . . . . . . . . . . . . . . . . . . . . . . . . . 35
Operational display . . . . . . . . . . . . . . . . . . . . . . . . . . 38
Operational safety . . . . . . . . . . . . . . . . . . . . . . . . . . . 10
Order code . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 13, 14
Orientation (vertical, horizontal) . . . . . . . . . . . . . . . . 19
Outlet runs . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 20
Output signal . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 130
Output variables . . . . . . . . . . . . . . . . . . . . . . . . . . . 130

P

Packaging disposal . . . . . . . . . . . . . . . . . . . . . . . . . . 17
Parameter settings
Administration (Submenu) . . . . . . . . . . . . . . . . . . 74
Advanced setup (Submenu) . . . . . . . . . . . . . . . . . 67
Analog inputs (Submenu) . . . . . . . . . . . . . . . . . . . 64
Communication (Submenu) . . . . . . . . . . . . . . . . . 63
Corrected volume flow calculation (Submenu) . . . . 68
Density adjustment (Wizard) . . . . . . . . . . . . . . . . 70
Device information (Submenu) . . . . . . . . . . . . . . 119
Diagnostics (Menu) . . . . . . . . . . . . . . . . . . . . . . 115
Low flow cut off (Wizard) . . . . . . . . . . . . . . . . . . . 65
Measured variables (Submenu) . . . . . . . . . . . . . . . 78
Medium selection (Submenu) . . . . . . . . . . . . . . . . 62
Partially filled pipe detection (Wizard) . . . . . . . . . 66
Sensor adjustment (Submenu) . . . . . . . . . . . . . . . 69
Setup (Menu) . . . . . . . . . . . . . . . . . . . . . . . . . . . . 59
Simulation (Submenu) . . . . . . . . . . . . . . . . . . . . . 75
System units (Submenu) . . . . . . . . . . . . . . . . . . . . 59
Totalizer (Submenu) . . . . . . . . . . . . . . . . . . . . . . . 80
Totalizer 1 to n (Submenu) . . . . . . . . . . . . . . . . . . 73
152

Proline Promass P 100 PROFIBUS DP

Totalizer handling (Submenu) . . . . . . . . . . . . . . . . 81
Web server (Submenu) . . . . . . . . . . . . . . . . . . . . . 44
Zero point adjustment (Submenu) . . . . . . . . . . . . . 72
Performance characteristics . . . . . . . . . . . . . . . . . . . 133
Performing density adjustment . . . . . . . . . . . . . . . . . 70
Pharmaceutical compatibility . . . . . . . . . . . . . . . . . . 145
Post-connection check . . . . . . . . . . . . . . . . . . . . . . . . 58
Post-connection check (checklist) . . . . . . . . . . . . . . . 33
Post-installation check . . . . . . . . . . . . . . . . . . . . . . . 58
Post-installation check (checklist) . . . . . . . . . . . . . . . 25
Potential equalization . . . . . . . . . . . . . . . . . . . . . . . . 31
Power consumption . . . . . . . . . . . . . . . . . . . . . . . . . 132
Power supply failure . . . . . . . . . . . . . . . . . . . . . . . . 132
Pressure Equipment Directive . . . . . . . . . . . . . . . . . 146
Pressure loss . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 139
Pressure-temperature ratings . . . . . . . . . . . . . . . . . 138
Process connections . . . . . . . . . . . . . . . . . . . . . . . . 142
Process variables
Calculated . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 129
Measured . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 129
Product safety . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 10
Protecting parameter settings . . . . . . . . . . . . . . . . . . 76

R

RCM mark . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 145
Read access . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 39
Reading measured values . . . . . . . . . . . . . . . . . . . . . . 78
Recalibration . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 122
Reference operating conditions . . . . . . . . . . . . . . . . 133
Registered trademarks . . . . . . . . . . . . . . . . . . . . . . . . . 8
Remote operation . . . . . . . . . . . . . . . . . . . . . . . . . . 143
Repair . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 123
Notes . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 123
Repair of a device . . . . . . . . . . . . . . . . . . . . . . . . . . 123
Repeatability . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 134
Replacement
Device components . . . . . . . . . . . . . . . . . . . . . . . 123
Requirements for personnel . . . . . . . . . . . . . . . . . . . . . 9
Response time . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 134
Return . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 123

S

Safety . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9
SensorMounting . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 23
Sensor heating . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 21
Sensor housing . . . . . . . . . . . . . . . . . . . . . . . . . . . . 138
Serial number . . . . . . . . . . . . . . . . . . . . . . . . . . . 13, 14
Setting the operating language . . . . . . . . . . . . . . . . . 58
Settings
Adapting the measuring device to the process
conditions . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 81
Administration . . . . . . . . . . . . . . . . . . . . . . . . . . . 74
Analog Input . . . . . . . . . . . . . . . . . . . . . . . . . . . . 64
Communication interface . . . . . . . . . . . . . . . . . . . 63
Low flow cut off . . . . . . . . . . . . . . . . . . . . . . . . . . 65
Medium . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 62
Operating language . . . . . . . . . . . . . . . . . . . . . . . 58
Partially filled pipe detection . . . . . . . . . . . . . . . . . 66
Endress+Hauser

Proline Promass P 100 PROFIBUS DP

Reset the totalizer . . . . . . . . . . . . . . . . . . . . . . . . 81
Resetting the device . . . . . . . . . . . . . . . . . . . . . . 118
Sensor adjustment . . . . . . . . . . . . . . . . . . . . . . . . 69
Simulation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 75
System units . . . . . . . . . . . . . . . . . . . . . . . . . . . . 59
Tag name . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 59
Totalizer . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 73
Totalizer reset . . . . . . . . . . . . . . . . . . . . . . . . . . . 81
SETTOT_MODETOT_TOTAL module . . . . . . . . . . . . . 55
SETTOT_TOTAL module . . . . . . . . . . . . . . . . . . . . . . 54
Signal on alarm . . . . . . . . . . . . . . . . . . . . . . . . . . . . 130
Software release . . . . . . . . . . . . . . . . . . . . . . . . . . . . 48
Spare part . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 123
Spare parts . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 123
Special connection instructions . . . . . . . . . . . . . . . . . 31
Special mounting instructions
Hygienic compatibility . . . . . . . . . . . . . . . . . . . . . 22
Standards and guidelines . . . . . . . . . . . . . . . . . . . . . 146
Status area
For operational display . . . . . . . . . . . . . . . . . . . . . 38
Status signals . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 85
Sterilization in place (SIP) . . . . . . . . . . . . . . . . . . . . 137
Storage conditions . . . . . . . . . . . . . . . . . . . . . . . . . . . 16
Storage temperature . . . . . . . . . . . . . . . . . . . . . . . . . 16
Storage temperature range . . . . . . . . . . . . . . . . . . . 137
Structure
Operating menu . . . . . . . . . . . . . . . . . . . . . . . . . . 36
Submenu
Administration . . . . . . . . . . . . . . . . . . . . . . . . . . . 74
Advanced setup . . . . . . . . . . . . . . . . . . . . . . . . . . 67
Analog inputs . . . . . . . . . . . . . . . . . . . . . . . . . . . 64
Calculated values . . . . . . . . . . . . . . . . . . . . . . . . . 67
Communication . . . . . . . . . . . . . . . . . . . . . . . . . . 63
Corrected volume flow calculation . . . . . . . . . . . . . 68
Device information . . . . . . . . . . . . . . . . . . . . . . . 119
Event list . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 116
Measured values . . . . . . . . . . . . . . . . . . . . . . . . . 78
Measured variables . . . . . . . . . . . . . . . . . . . . . . . 78
Medium selection . . . . . . . . . . . . . . . . . . . . . . . . . 62
Overview . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 37
Process variables . . . . . . . . . . . . . . . . . . . . . . . . . 67
Sensor adjustment . . . . . . . . . . . . . . . . . . . . . . . . 69
Simulation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 75
System units . . . . . . . . . . . . . . . . . . . . . . . . . . . . 59
Totalizer . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 80
Totalizer 1 to n . . . . . . . . . . . . . . . . . . . . . . . . . . . 73
Totalizer handling . . . . . . . . . . . . . . . . . . . . . . . . 81
Web server . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 44
Zero point adjustment . . . . . . . . . . . . . . . . . . . . . 72
Supply voltage . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 132
Surface roughness . . . . . . . . . . . . . . . . . . . . . . . . . . 143
Symbols
For communication . . . . . . . . . . . . . . . . . . . . . . . 38
For diagnostic behavior . . . . . . . . . . . . . . . . . . . . 38
For locking . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 38
For measured variable . . . . . . . . . . . . . . . . . . . . . 38
For measurement channel number . . . . . . . . . . . . 38
For status signal . . . . . . . . . . . . . . . . . . . . . . . . . . 38
In the status area of the local display . . . . . . . . . . . 38
Endress+Hauser

Index

System design
Measuring system . . . . . . . . . . . . . . . . . . . . . . . 128
see Measuring device design
System integration . . . . . . . . . . . . . . . . . . . . . . . . . . 48
System pressure . . . . . . . . . . . . . . . . . . . . . . . . . . . . 20

T

Technical data, overview . . . . . . . . . . . . . . . . . . . . . 128
Temperature range
Medium temperature . . . . . . . . . . . . . . . . . . . . . 137
Storage temperature . . . . . . . . . . . . . . . . . . . . . . . 16
Terminal assignment . . . . . . . . . . . . . . . . . . . . . . 27, 29
Terminals . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 132
Thermal insulation . . . . . . . . . . . . . . . . . . . . . . . . . . 20
Tool
For mounting . . . . . . . . . . . . . . . . . . . . . . . . . . . . 23
Transport . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 16
Tools
Electrical connection . . . . . . . . . . . . . . . . . . . . . . 26
TOTAL module . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 54
Totalizer
Configuration . . . . . . . . . . . . . . . . . . . . . . . . . . . . 73
Operation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 81
Reset . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 81
Transmitter
Connecting the signal cables . . . . . . . . . . . . . . . . . 29
Turning the display module . . . . . . . . . . . . . . . . . 24
Transporting the measuring device . . . . . . . . . . . . . . 16
Troubleshooting
General . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 83
TSE/BSE Certificate of Suitability . . . . . . . . . . . . . . . 145
Turning the display module . . . . . . . . . . . . . . . . . . . . 24

U

UKCA marking . . . . . . . . . . . . . . . . . . . . . . . . . . . . 145
Use of the measuring device
Borderline cases . . . . . . . . . . . . . . . . . . . . . . . . . . . 9
Incorrect use . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9
see Intended use
User interface
Current diagnostic event . . . . . . . . . . . . . . . . . . . 115
Previous diagnostic event . . . . . . . . . . . . . . . . . . 115
User roles . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 37
USP Class VI . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 145

V

Version data for the device . . . . . . . . . . . . . . . . . . . . 48
Vibration- and shock-resistance . . . . . . . . . . . . . . . . 137
Vibrations . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 22

W

W@M . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 122, 123
W@M Device Viewer . . . . . . . . . . . . . . . . . . . . . . . . 12
WeightSI units . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 140
Transport (notes) . . . . . . . . . . . . . . . . . . . . . . . . . 16
US units . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 140
WizardDefine access code . . . . . . . . . . . . . . . . . . . . . . . . 76
Density adjustment . . . . . . . . . . . . . . . . . . . . . . . 70
153

Index

Proline Promass P 100 PROFIBUS DP

Low flow cut off . . . . . . . . . . . . . . . . . . . . . . . . . . 65
Partially filled pipe detection . . . . . . . . . . . . . . . . . 66
Workplace safety . . . . . . . . . . . . . . . . . . . . . . . . . . . 10
Write access . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 39
Write protection
Via access code . . . . . . . . . . . . . . . . . . . . . . . . . . . 76
Via write protection switch . . . . . . . . . . . . . . . . . . 77
Write protection switch . . . . . . . . . . . . . . . . . . . . . . . 77

154

Endress+Hauser

*71601225*
71601225

www.addresses.endress.com