MCPD-8(Data sheet V2.0_01)

Data collector module formesytec psd+ system

MCPD-8 is a module which collects the data of up to eight MPSD-8+ or MSTD-16 modules, createsa full time stamp with 48bits width, buffers the position amplitude and timing data and sends it via
ethernet to a data acquisition PC.
It also transmits the configuration data between control PC and the peripheral modules. There is acommunication path via synchronisation line to the other MCPD-8 or MDLL modules in a setup, allowing to synchronise the time stamps between those data collectors, and transmit basic commands like start, stop and run number.

Features:•High rate capability up to 2 mega events/s permanently. Noadditional dead time up to the rate limit. Bursts with rates up
to 6.4 mega events/s can be stored in the main buffer.

•48 bit time stamp for all events with 100ns resolution(running 325 days).

•6 external digital inputs for connection of monitor countersor chopper signals. Can create fully time stamped events ofmultiple types which are fed to the event data stream.

•2 voltage inputs digitized by 12bit ADCs, sample rate
max. 100kHz. Data can be time stamped and sent to the datastream•

2 voltage outputs, voltage created by 12bit DACs. Can becontrolled via control PC.

•

2 digital outputs, controlled by PC•

Synchronisation bus, is a Lemo coax connection between
several MCPD-8 or MDLL modules. One MCPD-8 module
is configured as master module and synchronises all connected modules. Also basic signals like start, stop and a run
number is transmitted.
The master MCPD-8 itself can be synchronised to an externally applied 10 MHz clock (i.e. chopper time base)•PC Software: MCPD-8 and MDLL communicate with thecontrol and (if existing separately) data acquisition PCs usinga UDP based protocol. The open source data acquisition andcontrol software mesydaq/ qmesydaq, running on LinuxOS, is provided for data acquisition, online visualisation andsystem control.

mesytec GmbH & Co. KG
Wernher-von-Braun-Str. 1, D-85640 Putzbrunn, Germany
phone: +49 - 89 / 456007-30 fax: +49 - 89 / 456007-39
info@mesytec.com - www.mesytec.com

MCPD-8

Data collector modulefor mesytec PSD+ system

Functional Overview

fig. 5 : internal design of the MCPD-8 module

The internal design mainly consists of two blocks.:
The first is a programmable logic, which services theeight fast event buses, creates the time stamps,
accumulates all incoming data in a 1Mbyte FIFO(= 128kevents), and counts the external monitor
signals. As additional features the MCPD-8 has two
ADC inputs (12 bit) and can create periodic timestamped ADC events which are accumulated with theother events in the FIFO. This is for example useful toadd additional experiment parameters (temp, pressure,fields, …) to the data stream.
The programmable logic does not add any additional
dead time to the event bus. It can manage event bus
rates of up to 6.4MHz.

The second block is a fast 16bit microprocessor which
delivers an additional buffer of up to 1Mbyte and services the ethernet connection. It can deliver the theoretical maximum transmission data rate of the 100Mbit on
ethernet. This is possible due to a fast DMA unit which
directly dumps the list mode data from FIFO to theethernet controller without stopping the data accumulation. The maximum event transmission rate to theethernet will be around 2Mevents/s.
(6bytes per event: 10bit amplitude, 10bit position andtime stamp + 5% protocol overhead)
A system with 128 detector tubes (16 x MPSD-8,2 x MCPD-8) will process a peak rate of 12.8MHz(detector limit100 kHz per detector) and an averagerate of 4MHz.

mesytec GmbH & Co. KG
Wernher-von-Braun-Str. 1, D-85640 Putzbrunn, Germany
phone: +49 - 89 / 456007-30 fax: +49 - 89 / 456007-39
info@mesytec.com - www.mesytec.com

2/5MCPD-8

Data collector modulefor mesytec PSD+ system

Front panel elements

mesytec GmbH & Co. KG
Wernher-von-Braun-Str. 1, D-85640 Putzbrunn, Germany
phone: +49 - 89 / 456007-30 fax: +49 - 89 / 456007-39
info@mesytec.com - www.mesytec.com

3/5

MCPD-8

Data collector module
for mesytec PSD+ system

Rear panel elements

mesytec GmbH & Co. KG
Wernher-von-Braun-Str. 1, D-85640 Putzbrunn, Germany
phone: +49 - 89 / 456007-30 fax: +49 - 89 / 456007-39
info@mesytec.com - www.mesytec.com

4/5

MCPD-8

Data collector module
for mesytec PSD+ system

MCPD-8 data sheet
Power consumption:• P= 4W• U= +6V, 0.70A= -6V, 0.02A= +12V, 0.1A

(with oven stabilized quartz: up to 1A for 10s at power up)(optional: for DAC output only)

Event buses:• Maximum event bus data rate: up to 0.8MHz / bus• Event bus dead time per event: down to 1.2us• Signal level: 0 to +0.7V; Terminated inputs.
Ethernet:• 100Base-TX• Maximum Ethernet event rate to PC: 2MHz
6 Monitor, chopper inputs:• TTL input:• minimum High level 2.8V
• maximum low level: 0.8V
• Minimum pulse length: 200ns
2 Digital outputs:• 5V level, max. 10mA
2 DAC outputs:• 0 to 6V or ± 3V,(selected by jumpers)
max 30mA, protected
2 ADC inputs:• 0 to 9V or ± 4.5V
input resistance 6kΩ

(selected by jumpers)

Sync in / out:
• Provides and receives time reference and control signals• Oscillator: 1.25MHz (is multiplied by 8 in the receiver modules)
Time stamp generation:• Standard oscillator: stability 10-5
• Option: Oven stabilized oscillator: stability 10-7

mesytec GmbH & Co. KG
Wernher-von-Braun-Str. 1, D-85640 Putzbrunn, Germany
phone: +49 - 89 / 456007-30 fax: +49 - 89 / 456007-39
info@mesytec.com - www.mesytec.com

5/5