Battery Impedance Meter
BIM1000 Series

Easy & Reliable
Easy & Reliable
Battery Measurements

Maximumvoltage
measurement

1000V(BIM1100)

Battery Impedance Meter

BIM1000 Series●LineupBIM1030

BIM1030 WITH LAN

BIM1100

BIM1100 WITH LAN■ The best equipment for power battery production and inspection.
Ever-changing battery technology requires
batteries powering electric vehicles to havehigh voltage, high power and low impedance.
The Battery Impedance Meter, or BIM1000
Series, is capable of measuring up to 1000 Vof test voltage for simultaneous measurementsof both battery voltage and resistance at highspeeds. The BIM is the ideal equipment forpower battery development research and production tests.

● Maximum voltage measurement: 1000 V max. (BIM1100), 300 V max. (BIM1030)● Voltage measurement accuracy: ±(0.01 % of reading +3 digit)● Resistance measurement accuracy: ±(0.5 % of reading +5 digit)● Resistance measurement ranges: 3 mΩ/30 mΩ/300 mΩ/3 Ω● High resolution: Voltage 10 μV(6 V range), Resistance 0.1 μΩ(3 mΩ range)● Measurement frequency: 1 kHz ±0.2 Hz● Sampling speed(Resistance & voltage simultaneous measurements): 20 ms(FAST)● Zero Adjustment Function: Effective for decreasing measurement error			(If zero adjustment has been performed, “0 ADJ” is displayed)● Measurement logging(500 pairs) and collective transfer function● SIGNAL I/O, RS232C and USB as standard interface(BIM1030 WITH LAN and BIM1100 WITH LAN are standard equipped also with a LAN port)● New high visibility color display

Color liquid crystal display (LCD)

Comparator Functions

High visibility color monitor.
The resistance, voltage, upper and lower limits values
are displayed at a glance.

The comparator functions allows setting HIGH/LOW, andresistance and voltage can be simultaneously judged byindependent comparators. Judgment results are provided
on the display. External I/O is available for signal output.

Applications

For production line testing etc.

Unless specified otherwise, the specifications are for the following settings and conditions.
• The product is warmed up for at least 30 minutes.
• TYP: These are typical values that are representative of situations where the product operates in an environment
with an ambient temperature of 23 °C (73.4 °F). These values do not guarantee the performance of this product.
• setting: Indicates a setting.
• range: Indicates the rated value of each range.
• reading: Indicates a readout value.

■ Specifications■ Voltmeter (The range can be AUTO setting available)
Item■ Other functions

BIM1030 or BIM1030 WITH LAN / BIM1100 or BIM1100 WITH LAN

Rated input

BIM1030(WITH LAN): ±300 V / BIM1100(WITH LAN): ±1000 V

Range

6V

60 V

300 V
(BIM1030)

600 V(BIM1100)

1000 V(BIM1100)±6.30000 V±63.0000 V±315.000 V±630.000 V±1050.00 V

10 μV

100 μV

Maximum display value *1
Resolution
Accuracy *2

1 mV

10 mV

±(0.01 % of reading +3 digit)

Temperature coefficientApprox. 1 ms

Maximum display value *1
Resolution

3 mΩ

30 mΩ

300 mΩ

3Ω

3.1000 mΩ

31.000 mΩ

310.00 mΩ

3.1000 Ω

0.1 μΩ

1 μΩ

10 μΩ

100 μΩ

10 mA

1 mA

100 mA

Measurement frequency

1 kHz ±0.2 Hz

Accuracy *3±(0.05 %of reading +1 digit)/°C

Response time *4

Item

BIM1030 or BIM1030 WITH LAN / BIM1100 or BIM1100 WITH LAN

Sampling speed

FAST
50 Hz
60 Hz

20 ms

MEDIUMSLOW50 ms

160 ms

42 ms150 ms■ Judgment function
Item
Judgment method

BIM1030 or BIM1030 WITH LAN

BIM1100 or BIM1100 WITH LAN

Window comparator method. Judgment made with software.

Setting range

0.0000 Ω to 3.1000 Ω

Resolution

100 μΩ

Setting range

0.000 V to 315.000 V

0.00 V to 1050.00 V

Resolution

1 mV

10 mV

■ Interface
Item

BIM1030 or BIM1030 WITH LAN / BIM1100 or BIM1100 WITH LAN

RS232CD-SUB 9-pin connector, EIA-232-D compliant
Complies with USB Specification 2.0.
Data rate: 12 Mbps max. (Full Speed)USB

LAN *1Locks the key operation.
Zero adjustment of the voltmeter and resistance meter.
OFF setting available. Zero point clear function available.

Adjustment range

1000 digit

Measurement logging andcollective transfer function

Records up to 500 sets of measurement logs.
Logs can be read collectively.
Outputs an EOM signal from the SIGNAL I/O connector when ameasurement is completed.

HOLD

When the trigger source is set to INTERNAL, the signal is turned onafter a measurement is completed until the next measurement starts.
When the trigger source is set to EXTERNAL, the signal is turned onafter a measurement is completed until the next trigger is detected.

PULSE

Outputs a pulse when a measurement is completed.
Pulse width: 1 ms to 99 ms
Accuracy

ItemApprox. 2 ms

■ Sampling timeVoltage

Saves up to 100 sets of measurement conditions.

±(0.05 % of reading +0.5 digit)/°C*1. Displays OVER when the measurement range is exceeded.
*2. Within error ± 10 %.
*3. Add ±3 digit when the sampling speed is set to FAST and ±2 digit when the sampling speed is set to MID.
*4. The time for the product’s measurement circuit to stabilize when a probe in an open state isconnected to the DUT.

Resistance

Memory function

Complies with USBTMC Specification 1.0 and
USBTMC-USB488 Specification 1.0
IEEE 802.3, 100Base-Tx/10Base-T Ethernet, IPv4, RJ45,
Auto-negotiation supported but Auto MDI/MDI-X not supported.

SIGNAL I/O

D-SUB 25-pin connector.

*1. BIM1030 WITH LAN or BIM1100 WITH LAN only

Options■ Clip-type four-wire test lead
TL01-BIM■ Pin-type four-wire test lead
TL02-BIM±0.2 ms■ General specifications±(0.5 % of reading +5 digit)

Power supplyfrequency

The average count can be set between 2 and 99.
OFF setting available.

Four-terminal measurement method

Measured current *2

Temperature coefficient±0.2 ms

Average function

BIM1030 or BIM1030 WITH LAN / BIM1100 or BIM1100 WITH LANRange

0 to 9.999 s, OFFAccuracy

EOM function■ Resistance meter (The range can be AUTO setting available)
Item

Trigger delay

Zero adjustment*1. Displays OVER when the measurement range is exceeded.
*2. Add ±2 digit when the sampling speed is set to FAST or MID.
*3. The time for the product’s measurement circuit to stabilize when a probe in an open state isconnected to the DUT.

Measurement method

BIM1030 or BIM1030 WITH LAN / BIM1100 or BIM1100 WITH LAN
Select external trigger (EXTERNAL) or internal trigger (INTERNAL).

key lock±(0.001 % of reading +0.3 digit)/°C

Response time *3

Item
Trigger Function■ Zero adjustment toolOP01-BIM

BIM1030 or BIM1030 WITH LAN

Indoors, 2000 m or lessSpec
Temperature
guaranteed
Humidityrange

20 %rh to 80 %rh (no condensation)

Environment
Operating Temperature
range
Humidity
Storagerange
Powersupply

BIM1100 or BIM1100 WITH LAN

Installation location18 °C to 28 °C (-4 °F to 158 °F)
0 °C to 40 °C (32 °F to +122 °F)
20 %rh to 80 %rh (no condensation)

Temperature-10 °C to 60 °C (-4 °F to 158 °F)

Humidity

90 %rh or less (no condensation)

Input voltage range/
Input frequency range

85 Vac to 264 Vac (100 Vac to 240 Vac)/47 Hz to 63 Hz

Rated power

Isolation voltage
Insulation resistance
Between the AC LINE
and the chassis
Between all the meaWithsurement terminals andstandingthe chassis
voltage
Between all the measurement terminals andSIGNAL I/OExternal dimensions/ Weight

30 VA±300 V max±1000 V max

30 MΩ or more (500 Vdc)(between AC LINE and chassis)
1500 Vac for 1 minute, 10 mA or less2000 Vdc for 1 minute, 1 mA or less2000 Vdc for 1 minute, 1 mA or less214(8.43)W×80(3.15)H×300(11.81)D mm(inches)(Does not include protrusions)/ Approx. 3 kg (6.6 lbs)

Electromagnetic
compatibility (EMC) *1 *2

Complies with the requirements of the following directive and standards.
EMC Directive 2014/30/EU
EN 61326-1 (Class A *3), EN 55011 (Class A *3, Group 1 *4),
EN 61000-3-2, EN 61000-3-3

Safety *1

Complies with the requirements of the following directive and standards.
Low Voltage Directive 2014/35/EU *2
EN 61010-1 (Class I *5, Pollution Degree 2 *6),
EN 61010-2-030, EN 61010-031*1. Does not apply to specially ordered or modified products.
*2. Only on models that have CE/UKCA marking on the panel.
*3. This is a Class A instrument. This product is intended for use in an industrial environment. This product may cause interference if used in residential areas. Such use must be avoided unless the user
takes special measures to reduce electromagnetic emissions to prevent interference to the receptionof radio and television broadcasts.
*4. This is a Group 1 instrument. This product does not generate and/or use intentionally radio-frequency energy, in the form of electromagnetic radiation, inductive and/or capacitive coupling, for thetreatment of material or inspection/analysis purpose.
*5. This is a Class I instrument. Be sure to ground this product's protective conductor terminal.The safetyof this product is guaranteed only when the product is properly grounded.
*6. Pollution is addition of foreign matter (solid, liquid or gaseous) that may produce a reduction ofdielectric strength or surface resistivity. Pollution Degree 2 assumes that only non-conductive pollution will occur except for an occasional temporary conductivity caused by condensation.

■Distributor:

1-1-3, Higashiyamata, Tsuzuki-ku, Yokohama, Kanagawa, 224-0023, Japan
Phone:(+81)45-593-0200, Facsimile:(+81)45-593-7591, https://global.kikusui.co.jp/■All products contained in this catalogue are equipment and devices that are premised on use under the supervisionof qualified personnel, and are not designed or produced for home-use or use by general consumers. ■Specifications,design and so forth are subject to change without prior notice to improve the quality. ■Product names and prices are
subject to change and production may be discontinued when necessary. ■Product names, company names and brandnames contained in this catalogue represent the respective registered trade name or trade mark. ■Colors, texturesand so forth of photographs shown in this catalogue may differ from actual products due to a limited fidelity in printing.
■Although every effort has been made to provide the information as accurate as possible for this catalogue, certain
details have unavoidably been omitted due to limitations in space. ■If you find any misprints or errors in this catalogue,
it would be appreciated if you would inform us. ■Please contact our distributors to confirm specifications, price,accessories or anything that may be unclear when placing an order or concluding a purchasing agreement.

Issue:May.2023

202305PDFEC31