【ACCESSORIES】※Included for use when
adjusting for wear of
micrometer screw.

Probe tips

Reference Line

0-Point Adjustment Screw

Ratchet Stop

(Ring gauge calibration standard isnot included.)(Constant Force Mechanism)

READING THE SCALE(e.g.) MCA-1620S

Thimble Scale

Vernier Scale②Reference Line past
the 6２０１６０８６４２０③Vernier Scale aligns
at 6 on the Thimble

Reference LineSleeve Scale０■Resolution = 0.005mm
Sleeve Scale(e.g.) MC-2025IPS

Thimble Scale

Reference Line２５２０■Resolution = 0.002mm

４５５０1０５①Sleeve reading greater than 16① Sleeve Scale Reading :１６．
０② Thimble Scale Reading :．
０６③ Vernier Scale Reading :．
００６

Reading :１６．
０６６（mm）

Read the Thimble Scale at the Reference LineSleeve Scale Reading ２０．
０Thimble Scale Reading．
０１５
Reading ２０．
０１５（mm）

CAUTION● Always wipe measuring surfaces with a cloth to remove any dirt before use.
● Do not turn spindle more than 0.5mm above upper measurement range.
● If probe tips do not return after Thimble is loosened, please use hand in dust free glove to push in.
● This is a precision instrument, handle with care. Do not drop or subject to excessive forces.
● Please do not disassemble or modify.
● Keep instrument away from direct sunlight or high temperatures such as in a car, or near a stove or heat source.
● Remove dust and cutting chips after use and apply rust preventative oil to metallic surfaces. Please keep out of reach of children.
● Please do not use organic solvents to clean the body of the instrument.
● Keep instrument away from water and oil.
● Only use as directed. Improper use may cause accident or injury.

Ratchet Stop

Adjust Thimble to matchvalue of ring gauge

Ring gauge
calibration standard(e.g.) 20.005mm

3) Tighten the screw above the Ratchet Stop, then loosen the0-Point Adjustment Screw. Zero-setting is now complete.

MICROMETER SCREW WEAR ADJUSTMENTOPERATION① Remove any grease and dust
from the measuring surfaces witha clean cloth, such as a lenscleaning cloth. Make sure no
foreign objects are adhered tothe surfaces.
② Insert the measurement head intoobject to be measured and turnSpindle until the measuring
surfaces make contact with theinside diameter. Continue turning
until Ratchet Stop “clicks” 1~3times.
③ Read the measured value
following the “Reading the Scale”
procedure above.

“Click, Click”

Wear on the micrometer screw in the sleeve will affect accuracyof the measurement. Please adjust using the key provide asrequired.
① Remove the Thimble by rotating as shown.
② Insert the Key into the notches in the exposed brass nut.
③ Rotate the Key in the clockwise direction a little at a time.
(Over-tightening will make the Thimble hard to turn)Tighten
２５・Key５LoosenThimble moves again

４５０RodDetermine if the scalereading matches thering gauge size.

5２０－２５ｍｍ

0-Point
Adjustment Screw

2) Using the Allen Wrench, loosen
the screw above the Ratchet Stop
until the Thimble is able to moveagain. Turn the Thimble to adjust
the reading to the proper setting.

Key４５２０－２５ｍｍ

０５Inser t

２０２５・Allen Wrench

Tighten

Locking the Thimble

２０Thimble

Allen Wrench

Allen Wrench

45Sleeve

Gauge Head0-Point
Adjustment Screw

Sleeve0PART IDENTIFICATION

1) Tighten the 0-Point Adjustment
Screw using the supplied AllenWrench to prevent the Thimblefrom turning.

Thimble２０－２５ｍｍ● Storage Temperature Range : 0〜50℃

MCA-0608S
MCA-0810S
MCA-10125S
MCA-12516S
MCA-1620S
MC-2025IPS
MC-2535IPS
MC-3550IPS

5● Operating Temperature Range : 15〜40℃

Ratchet Stop

“Click, Click”

45● Maximum permissible error : 4μm

Measurement Resolution Measuring Surface Weight(mm)
(mm)
(g)
Range
Material
6 〜 895
8 〜 10105
Alloy Tool Steel
0.002
10 〜 12.5
115
12.5 〜 16140
16 〜 20155
20 〜 25
Carbide Tip
3050.005
25 〜 35325
35 〜 50465③ Determine if the micrometer reading from the Spindle Scaleand Thimble matches the size of the ring gauge calibrationstandard.
④ If they do not match, without moving Thimble from position
set in step ② above, perform the following procedure.

0Model No.

２５２０SPECIFICATIONS

２０－２５ｍｍ● 3 Point measurement gives high accuracy measurement of internal diameter.

５FEATURES

０Thank you for purchasing the S-Line THREE-Point Internal Micrometer.
Please read this manual thoroughly before use for proper operation. Ring gauge calibration standard is not included forzero-setting, please prepare appropriate ring gauge as required by the instrumentʼs measurement range.

※Use ring gauge appropriate for instrument measuring range.

① Prepare instrument and ring gauge for calibration by cleaningmeasurement head and ring gauge inner diameter withappropriate cleaner.
② Insert the measurement head into the ring gauge and turn
Spindle until the measuring surfaces make contact with theinside of the ring gauge. Continue turning until Ratchet Stop
“clicks” 1~3 times.

４５

Model No. : MCA-*****S / MC-****IPS <combined>

ZERO-POINT SETTING２５２０INSTRUCTION MANUAL

２５２０

S-LINE THREE-POINT INTERNAL MICROMETERSleeve

NutThimbleRemove

Object to be measured④ Replace Thimble and then perform the zero-point setting
procedure described above.

TROUBLESHOOTINGERROR CONDITION

CORRECTIVE ACTION

Measured value is wrong●Probe tips do not return●Clean measuring surfaces with cloth.
Check to insure zero-point is correct. Please reset zero-point.
● If micrometer nut seems worn, please adjust using the above
procedure.
●Please push in by hand in dust free glove.

※If issue does not resolve, or if you have any questions, please contact distributor or place of purchase.
※Please note, manufacturer is unable to respond to inquires or provide service directly.
Please contact distributor or place of purchase.

Niigata seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

４５※付属のキーはマイクロメータナットが磨耗した時の調整に使用します。

０点調整ねじ基準線

リングゲージ寸法値と数値が一致しているか確認してください。

基準リングゲージ

マイクロメータナットが磨耗した時の調整

基準線
２５２００８６４２００１６５２０

シンブルの目盛０1０① スリーブの読み：１６．０② シンブルの読み：．０６③ 副尺の読み：．００６

読み：１６．０６６（mm）

５

基準線とシンブルの目盛が合った所を読む②測定部を被測定物の内径に入れ、ラチェットストップを持って回します。測定子と被測定物の内径が接触し、ラチェットストップが『カチカチ』と１〜３回鳴ったところで停止してください。
③表面『目盛の読み方』に従って、スリーブとシンブルの目盛を読み取ってください。

スリーブの読み：２０．０
シンブルの読み：．０１５
読み：２０．０１５（mm）

ご注意● ご使用の前に、必ず測定子を含めた測定部を脱脂し、グラスペーパなどで拭き取って汚れを除去してください。
● 測定範囲の上限を０.５mm以上上回ってシンブルを回さないでください。

スリーブの内部にあるマイクロメータナットが磨耗して、測定の精度が低くなった場合は付属のキーを使って調整してください。

カチカチ①シンブルを回して外してください。

②真鍮製のナットの切り欠き部分にキーを入れます。
③キーを時計回りに回し少しずつ締めます。
（締めすぎるとシンブルが回りにくくなります。）
締める

５

スリーブ

トラブルと対応
トラブルの内容
トラブルの内容

対応策

測定面の汚れをグラスペーパなどで拭き取ってください。
● 原点が合っているか確認し、設定し直してください。
● マイクロメータナットが磨耗しているようであれば、調整を行ってください。

● 使用後は柔らかい布などでゴミや切粉などを取り除き、防錆油を塗布してお子様の手の届かない場所へ保管してください。

測定子が戻らない●● 水や油のかかるところで使用しないでください。

※症状が改善されない場合や不明な点がございましたらお買い上げの販売店、または発売元までご連絡ください。
※お問い合わせや、ご連絡が無いまま直接発売元に修理品などを送付されても処理、対応ができない場合があります
のでご了承ください。

● 測定部の分解・改造はしないでください。

● 用途以外のご使用は、事故やけがの原因となりますので絶対におやめください。

外す④シンブルを組み直し、上記手順に従って原点を再調整してください。

被測定物● 精密機器ですので、落としたりぶつけたり、強い力を加えたりしないでください。

● 本体のお手入れには有機溶剤を使用しないでください。

シンブル

ナット

●

● 直射日光の当たる場所や炎天下の車の中、火のそばなどでの使用、保管はしないでください。

４５０

測定値が間違っている● シンブルをゆるめても測定子が戻らない場合は、手袋等を着用し、手で押し込んでください。

キー２０－２５ｍｍ

入れる①スリーブの16を超えている

４５①測定子を含めた測定部を脱脂し、グラスペーパなどで拭き取って異物が付着していない状態にしてください。

２５スリーブの目盛例）ＭＣ-２０２５ＩＰＳ

5■最小読取値 ０.００５mm

0②基準線がシンブルの6を超えている③シンブルと合った所の副尺目盛が６5３）ラチェットストップの後ろにあるねじを締め、０点調整ねじをゆるめて、原点設定の完了です。

（例）２０．
００５mm

２０－２５ｍｍ

シンブルの目盛45

リングゲージの寸法値と数値を合わせる

45例）ＭＣＡ-１６２０Ｓ

ラチェット
ストップ

シンブルが再び動く

使用方法

副尺の目盛

基準線
スリーブの目盛０点調整ねじゆるめる（リングゲージは付属しておりません。）

目盛の読み方

■最小読取値 ０.００２mm

六角棒レンチ２）六角棒レンチでラチェット
ストップ の後ろのねじをシンブルが再び動く程度ゆるめ、 リングゲージの寸法値とスリーブ・シンブルの目盛の数値を一致させます。

２５２０

測定子

スリーブ・キー

０５ロッド

シンブル

5２５２０－２５ｍｍ【付属品】

・六角棒レンチ

締める

シンブルを固定する

45

ラチェットストップ（定圧装置）

シンブル

カチカチ

六角棒レンチ0スリーブ

測定部

２０各部の名称

ラチェットストップ

0MCA-0608S
MCA-0810S
MCA-10125S
MCA-12516S
MCA-1620S
MC-2025IPS
MC-2535IPS
MC-3550IPS●保管温度範囲：0〜50℃

０点調整ねじ２０●使用温度範囲：15〜40℃

測定範囲（mm）最小読取値（mm） 測定子材質
質量（ｇ）
6 〜 895
8 〜 10105
合金工具鋼鋼材
10 〜 12.5
115
0.002
12.5 〜 16140
16 〜 20155
20 〜 25305
超硬チップ
25 〜 35325
0.005
35 〜 50465番5品１）シンブルが回らなくなるまで、０点調整ねじを付属の六角棒レンチで締めます。

0●最大許容誤差：4μm④一致していない場合は、②の状態からシンブルを動かさないように、以下の手順で０点を合わせてください。

２０－２５ｍｍ様45仕５●内径を３点で測定するため、より精度の高い測定が可能です。

０長②測定部をリングゲージの内径に入れ、ラチェットストップを回します。測定子とリングゲージの内径が接触し、ラチェットストップが『カチカチ』と１〜３回鳴ったところで停止してください。

４５

特③リングゲージに表示されている寸法値と、スリーブ・シンブルの目盛が指している数値が一致していれば、０点が合っています。

①測定子を含めた測定部と、お手持ちの基準リングゲージの内径をパーツクリーナーなどの洗浄剤で洗浄します。

２５２０

この度はＳライン三点マイクロメータシリーズをお買上げいただきありがとうございます。
ご使用前に、本書をよくお読みいただき正しくお使いくださいますようお願い申し上げます。
基準のリングゲージは付属されておりませんので、原点の調整には測定範囲内のリングゲージをご用意ください。

※測定範囲内のリングゲージをご用意ください。

２５２０

品番：MCA- *****S／MC- ****IPS〈兼用〉

原点の調整

２５２０

Ｓライン三点マイクロメータ

取扱説明 書

手袋等を着用し、手で押し込んでください。

発売元〒955-0055 新潟県三条市塚野目5丁目3番14号(0256)33-5502(代) FAX(0256)33-5528

URL http://www.niigataseiki.co.jp

I318-K1 2204