DIGITAL LONG CALIPER

INSTRUCTION MANUAL
Model：Ｄ-５００／６００／１０００／１５００／２０００ ＜comb.＞

Thank you for purchasing Digital Long Calipers.
Before use, please read this document thoroughly for proper operation.
3

PARTS ID.

11

2

5

4­2

7

6

4­1
pre ＋

pre －

ZERO

13

10

OFF

ON

9

8

12

1

1. Outside Meas. Surfaces
2. Step Meas. Surfaces
3. Inside Meas. Surfaces
4. Lock Screws
5. LCD Display
6. Scale
7. Beam

8. Battery Cover
9. [ON/OFF] Button
10. [ZERO] Button
11. Preset Adj. Buttons
12. Fine Adjust Feed
13. Slide

BATTERY REPLACEMENTNOTICES

１． Slide the Cover in direction of arrow ▽ to Open.
２． Insert Battery with the (＋）side outward.
３． Replace Battery Cover.

・ Before use, please wipe off corrosion protectant
using a soft, oiled cloth.
・ This is a precision instrument, handle with care. Do
not drop or subject to excessive force.
・ Do not damage scale or scratch instrument. Do not
apply ID numbers with engraver.
・ Keep away from direct sunlight. Do not store
instrument in extreme temperatures.

CAUTION Only use CR2032 coin type Lithium Battery.

OPERATION＜Setting the Origin＞
Setting the Origin (Zero-Point) for absolute measurements.
Note, the Zero-Point is stored in memory, even when power isturned OFF so there is no need to set each time. When battery is
replaced the Zero-Point is will need to be reset.
１． Turn power ON.
２． Close the Caliper Jaws and press the [ZERO] Button.
３． The Display will read "0.00mm" or, "‐0.00mm".
※ If Display does not show zero, please use the following
procedure:① Press and hold the [ZERO] Button, and adjust the display
to zero by pressing and holding the Preset Adj. Buttons.
※ When “SET” on the Display blinks, the [ZERO] Button
can be released as you continue the adjustment.
② Press the [ZERO] Button.
The Zero-Point is now set.
＜Preset Function＞
User can set the Preset Value, and all measurements will be
relative to that value.
1. Press and hold the [ZERO] Button, and adjust the display to thedesired value by pressing and holding the Preset Adj. Buttons.
※ When “SET” on the Display blinks, the [ZERO] Button can bereleased as you continue the adjustment.
2. Press the [ZERO] Button.
※ Preset Value is stored in memory when power is turned off.
To change the value, or to reset to zero, use the above
procedure.
＜Fine Adjust Feed＞
１. Tighten the Fine Adj. Lock Screw (4-2), and loosen the Slide
Lock Screw (4-1).
２. Rotate Fine Adjust Feed Wheel to the desired value.

TROUBLESHOOTING
Q. Nothing is displayed on LCD, or the numbers areblinking.
A. The battery may be running low.
Q. With power on, the number on the LCD does notchange as Slide is moved.
A. Please reset by removing battery for about 30sec.,
and then replacing.
Q. Nothing happens when buttons are pressed.
A. Is the battery inserted properly?
Q. The measurement on the LCD is incorrect.
A. If the Slide is moved to quickly, the reading may beincorrect. Please do not move the slide too quickly.

SPECIFICATIONS・ Resolution:
・ I nst. Error:

・ Display:・ Power:

0.01mm
D-500････±0.05mm
D- 600････±0.08mm
D-1000･･･±0.10mm
D-1500･･･±0.14mm
D-2000･･･±0.20mm
D-500/600･･･････････5 char. LCD
D-1000/1500/2000･･･6 char. LCD
CR2032 Coin type Lithium Battery

Niigata seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

I119-K1

1106