Instruction Manual Model
D-***IP67S<comb.>
No.

Precision
Measuring Tool

DIGITAL S­LINE CALIPER

Thank you for purchasing Niigata Seiki Digital S-Line Caliper.
This caliper is IP-67 rated for water and dust resistance, and can be used in applications where it will be exposed to water,
cutting fluid, lubricating oil, etc.
● For safe and proper use, please read this instruction manual before use and follow the procedures described.
● Please keep manual where it is accessible to user for future reference.

● Keep this manual with the instrument if transferred or leased to a third party.

For inquiries about this product, please contact dealer or Niigata Seiki at the address listed on the following page.

■ PARTS IDENTIFICATION ・ FUNCTION⑬ Battery Cover Tool

⑩ LCD Display⑨ Lock Screw⑤ Inside Meas.
Surfaces

MICROMETER MCD-***IP67SCALIPER D-***IP67S④ Inside Jaws

⑥ Depth Bar③ Step Meas.
Surfaces

IP67mm

⑦ Depth Meas.
Surfaces

SWISS MADE

② Outside Jaws

MODESET⑪ Battery Cover⑫[SET]Button① Outside Meas.
Surfaces① Outside Meas. Surfaces ·· Position object to be measured
between these surfaces.
② Outside Jaws ·················· For measuring outside dimensions.
③ Step Meas. Surfaces ······ Measures height of steps.
④ Inside Jaws ····················· For measuring inside dimensions.
⑤ Inside Meas. Surfaces ····· Position surfaces inside object tobe measured.

SAFETY PRECAUTIONS⑧ Scale⑥ Depth Bar ················ For measuring depth.
⑦ Depth Meas. Surf. ···· Depth is measured between these twosurfaces.
⑧ Scale ························ Supports LCD Display Unit.
⑨ lock Screw ··············· Tighten to prevent the slide and display
from moving.
⑩ LCD Display ············· Displays digital measurement.
⑪ Battery Cover ··········· Open to access Battery.
⑫ [SET]Button ············ Power ON/OFF, Set 0-point, Sleep Mode.
⑬ Battery Cover Tool ········· Used to open and close Battery Cover ⑪.

Please Observe

To prevent harm to yourself or others, and to prevent damage toproperty, always follow the procedures marked by the following symbols.

WARNING:※D­150IP67S・D­200IP67Sonly

Denotes a prohibition –You MUST NOT do

Denotes a requirement –You MUST do

Indicates risk of personal injury or property damage if not followed.

Read the manual and follow the directions.

・Use of product other than as described in this manual may
cause accident.

Use only for measuring.

・Use for any purpose other than measuring may damage orwear the instrument. Improper use may also cause accident.

Handle with Care.
・ Do not drop or subject instrument to excessive shock. Do notplace under heavy objects. Improper handling may
cause damage or poor accuracy.
・ Do not scratch instrument, for example by writing ID number.

Avoid using in high electric fields, such as nearfluorescent lights or switching power supplies.
・ Incorrect value may be displayed on LCD.

CALIBRATION●To maintain instrument accuracy, periodic calibration isrecommended.

Wear of measuring surfaces from repeated use may affect accuracy
and periodic accuracy checks should be performed.

Use and store only in environment meeting the following
conditions:● Dry location protected from rain, water, and oils.

● Location that is not excessively hot.
● Location protected from use by children and unauthorizedpeople.
・ Use in places contrary to the above may cause damage tothe product resulting in accident or injury.

Jaw Tips Are Sharp—Handle With Care.
・Careless handling may cause injury.
Do not disassemble or modify.
・Disassembly may damage product or cause accident.

For repair, please contact distributor or place of purchase.

Do not write on instrument with electric engraver.
・ Do not engrave instrument body or jaw, such as an ID number
with engraver or electric pen.

Outside Japan,
Please contact distributor or place of purchase toinquire about calibration services.

INSERTING ・ REPLACING BATTERY② Remove the Cover.

① Unscrew the cover using theBattery cover Tool, side marked「CALIPER D-***IP67S」
Battery
Cover Tool

Battery
CoverTurnSWISS

M

ADE

す外SET

Battery CoverCR2032 Battery③ Insert the new battery with the ＋ side outward.
④ Replace the cover using the Battery Cover Tool.
NoticeBatt.

Contamination of surfaces may cause measurement error.
● Allow the caliper and item to be measured time to reach same temperature.
A temperature difference between the object to be measured andthe caliper may cause measurement error. Allow enough time forthe temperatures to equilibrate.
The SET button performs the following functions.

① Press and hold for 2s.・・・Display is set to 0
② Press and hold for 2s.・・・Sleep Mode

(Resume by moving Jaw or pressing any button)(Turn power ON by pressing any button two times)● Use care to avoid damaging terminals when inserting battery● The ＋ terminals may be broken if battery is pressed straightdown.
● Insert battery at angle, with the ＋ side outward as shown.

R203● Clean off any rust inhibitor or debris from measurement and sliding surfaces.

③ Press and hold for 2s.・・・Power OFF

Use only CR2032 type battery＋ Term.

Pressed Down

SlideIn＋ Term.

■Absolute Measurement Mode

① Turn power ON by pressing any button two times.
② With jaws closed, press and hold the [SET] Button for 2 sec.
③ When “0.00mm” is displayed on LCD, Caliper is ready to startmeasurement.

■Comparison Measure Mode

2CBefore use please make the following preparations.

IP67■ SET BUTTON

CR2032MODE

HOW TO USE

Batt.

TROUBLESHOOTING
Ｑ．Nothing is displayed on LCD.
Ａ．The battery may be running low, please replace battery witha new one.

Ｑ．With power on, the number on the LCD does not change as
Display Unit is moved.

Ａ．Please reset by removing battery for about 30sec., and thenreplacing.
Ｑ．No response when a button is pressed.
Ａ．Make sure that the battery is correctly installed.

①Turn power ON by pressing any button two times.
② With jaws set on reference part, press and hold the [SET]
Button for 2 sec.
③ When “0.00mm” is displayed on LCD, Caliper is ready to startmeasurement.

IP CODE SUMMARY・IP rating for electricalequipment defined byIEC standard forwater and solid bodyprotection.
Description

1st Digit

６Solid IngressProtection

2nd Digit

７Description
Liquid IngressProtection

ＩＰ６７
2nd Digit: Water proof rating
1st Digit: Direct contactprotection rating for ingressof solid foreign object
Degree of Protection

Test method

No ingress of dust allowed.
Dust tight.

Complete protection
from dust intrusion

Degree of Protection

Test method

No harmful effect when
immersed in water forspecified period and pressure.

Submersion for 30min.
at depth of 1m (fordevices of height lessthan 850mm)

higher for other than outside measurements
SPECIFICATIONS ※0.02mm※Not including quantization error (± 1 count)

AFTER USE CARE ・ STORAGE● After use, wipe off any dirt or oil from measuring surfaces, sliding
surfaces, and exterior using a dry cloth and perform rust prevention
treatment. For rust prevention, lightly apply anti-rust oil, or place inrust preventive bag.
● Store with Jaws slightly open.
If stored with Jaws closed tight, expansion due to temperaturechanges may cause excessive force which will cause inaccuracy.

● Store in a cool, dry, and dark location in the provided case.
Keep out of direct sunlight and moisture, and please keep securefrom unauthorized personnel.
● Remove battery if Caliper will not be used for a long period of time.

Model No.

D-150IP67S
D-200IP67S
D-300IP67S

Range(mm)0 -150
0 -2000 -300

※Inst. Error(mm)±0.02

Mass (g)±0.03170200270● Resolution: . . . . . . . . . . . . . . . . . . . 0.01mm● Display: . . . . . . . . . . . . . . . . . . . . . . 5 Digit LCD● Auto Off: . . . . . . . . . . . . . . . . . . . . . ~ 20 min.
● Maximum Response Speed: . . . . . . 2.5m/s● Operating Temperature Range: . . . 5°C to 40°C
● Power: . . . . . . . . . . . . . . . . . . . . . CR2032 Lithium Battery (1x)
● Battery Life: . . . . . . . . . . . . . . . . . . . approx. 1.5 years in normal use● Accessory: . . . . . . . . . . . . . . . . . . CR2032 Lithium Battery (for test)
Battery Cover ToolSERVICE● If Instrument is not working properly, or if you have any questions, please contact
distributor or place of purchase.
● Please note, manufacturer is unable to respond to inquires or provide service
directly. Please contact distributor or place of purchase.

Niigata Seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055+81-256-33-5522

FAX : +81-256-33-5518

MAIL: intl.sales@niigataseiki.co.jp
URL: http://www.niigataseiki.co.jp

E125­K519102000