Digit Multimeter
DME1600

61/2 digit resolution, Essential device of "Electronic Measurement"
Supporting basic measurement with variety of options
D I G I T A L

M U L T I M E T E R

DME1600
The DME1600 is a digital multi-meter with a resolution of 6 1/2
digit. It can be performed up to 2000 times per second at thesetting condition of 4 1/2 digit as fastest measurement, and it
can measures 50 times per second when it is set for the 6 1/2
digit. The DME1600 offers fully function of measurement for thevoltage, current, resistance, frequency and temperature which
can be used various application of measurement and evaluationin design, development and debugging of electronics devices.
The DME1600 provides USB and GPIB interface* as standard
feature for automated measurement besides manual operation.
Furthermore, the DME1600 offers wide range of options suchas 20-channel multi-point scanner card supporting the basicmeasurement.

● DME1600● DME1600GC (with GPIB)■ Resolution : 6 1/2 digit■ Display : 5 x 7 dot matrix VFD, dual display with 3-color■ Basic measurement function

DC voltage : 0.1 V, 1 V, 10 V, 100 V, 1000 V
AC voltage : 0.1 V, 1 V, 10 V, 100 V, 750 V
DC current : 10 mA, 100 mA, 1 A, 3 A
AC current : 1 A, 3 A
2-wire / 4-wire resistance : 100 Ω, 1 kΩ, 10 kΩ, 100 kΩ, 1 MΩ, 10 MΩ, 100 MΩ
Frequency : 3 Hz to 300 kHz
Continuity test
Diode test
Temperature test
■ Built-in USB Interface (GPIB Interface*: selected model)*Model with GPIB Interface : DME1600GC

Options
20-channel multi-point scanner card
[DME1600-OPT09]

Kelvin probe(for 4-wire resistance measurement)[DME1600-OPT07]

Thermocouple adapter[DME1600-OPT02]

4-wire test lead
[DME1600-OPT08]

K type thermocouple cable[DME1600-OPT11]

10-channel multi-point scanner card
[DME1600-OPT01]
10-channel thermocouple multi-pointscanner card
[DME1600-OPT12]

Specifications
DC Characteristics

AC Characteristics

AccuracyAccuracy• ±(% of reading + % of range)• The specifications are for the following conditions: 6 1/2 digit resolution, minimum two-hour warm up, and auto
trigger mode.
• Use the null function for the 2-wire / 4-wire resistance measurement method.

• ±(% of reading + % of range)• The specifications are for the following conditions: 6 1/2 digit resolution, minimum two-hour warm up, and slow
AC filter (3 Hz to 300 kHz bandwidth).
• Measured using a sine wave input whose amplitude is greater than 5% of range.

DC voltage
Range
100.0000 mV
1.000000 V
10.00000 V
100.0000 V
1000.000 V
DC current
Range
10.00000 mA
100.0000 mA
1.000000 A
3.000000 A
Resistance
Range
100.0000 Ω
1.000000 kΩ
10.00000 kΩ
100.0000 kΩ
1.000000 MΩ
10.00000 MΩ
100.0000 MΩ
Diode test
Range
1.0000 V
Continuity test
Range
1 kΩAC voltage (true rms value)
Range

Resolution
0.1 μV
1.0 μV
10 μV
100 μV
1 mV

Input resistance> 10 GΩ> 10 GΩ> 10 GΩ
10 MΩ
10 MΩ

1 year (23 °C ± 5 °C)
0.0050+0.0035
0.0040+0.0007
0.0035+0.0005
0.0045+0.0006
0.0045+0.0010

Resolution
10 nA
100 nA
1 μA
10 μA

Input resistance
5.1 Ω
5.1 Ω
0.1 Ω0.1 Ω

1 year (23 °C ± 5 °C)
0.050+0.020
0.050+0.005
0.100+0.010
0.120+0.020

Resolution
100 μΩ
1 mΩ
10 mΩ
100 mΩ
1Ω
10 Ω
100 Ω

Input resistance
1 mA
1 mA
100 μA
10 μA
5 μA
500 nA
500 nA||10 MΩ

1 year (23 °C ± 5 °C)
0.010+0.004
0.010+0.001
0.010+0.001
0.010+0.001
0.010+0.001
0.040+0.001
0.800+0.010

Resolution
10 μV

Test current
1 mA

1 year (23 °C ± 5 °C)
0.010+0.020

Resolution
10 mΩ

Test current
1 mA

1 year (23 °C ± 5 °C)
0.010+0.030

100.0000 mV

1.000000 V to 750.000V

AC current (true rms value)
Range

Item
DC voltage measurement :Overrange
DC voltage measurement :Input bias current
DC voltage measurement :Input voltage protection
DC current measurement :Overrange
Resistance measurement :Maximum test lead resistanceResistance measurement :Input voltage protection

Specification
Permits voltages that are up to 20 % over the range except
when the 1000 V range is in use
Less than 30 pA (at 25 °C)
1000 V for all ranges
Permits currents that are up to 20 % over the range except
when the 3 A range is in use
10 Ω (100 Ω range),
100 Ω (1 kΩ range),
1 kΩ (other ranges)
1000 V for all ranges

Frequency and period characteristicsAccuracy• ±(% of reading)• The specifications are for the following conditions: 6 1/2 digit resolution and minimumtwo-hour warm up.

RangeFrequency
3 Hz to 5 Hz
5 Hz to 10 Hz
10 Hz to 40 Hz
40 Hz to 300 kHz

100 mVrms to 750 Vrms

1 year (23 °C ± 5 °C)
0.10
0.05
0.03
0.01

Measurement characteristics
ItemSpecification
Permits voltages that are up to 20 % over the range except
when the 750 Vrms range is in use
The maximum frequency for the 750 Vrms range is 100 kHz.

Overrange
Measurement frequency

0.1 μV

1.0 μV to 1 mVResolution

1.000000 A

1 μA

3.000000 A

10 μA

Frequency
3 Hz to 5 Hz
5 Hz to 10 Hz
10 Hz to 20 kHz
20 kHz to 50 kHz
50 kHz to 100 kHz
100 kHz to 300 kHz
3 Hz to 5 Hz
5 Hz to 10 Hz
10 Hz to 20 kHz
20 kHz to 50 kHz
50 kHz to 100 kHz
100 kHz to 300 kHz

1 year (23 °C ± 5 °C)
1.00+0.04
0.35+0.04
0.06+0.04
0.12+0.05
0.60+0.08
4.00+0.50
1.00+0.03
0.35+0.03
0.06+0.03
0.12+0.05
0.60+0.08
4.00+0.50

Frequency
3 Hz to 5 Hz
5 Hz to 10 Hz
10 Hz to 5 kHz
3 Hz to 5 Hz
5 Hz to 10 Hz
10 Hz to 5 kHz

1 year (23 °C ± 5 °C)
1.00+0.04
0.30+0.04
0.10+0.04
1.10+0.06
0.35+0.06
0.15+0.06

Measurement characteristics
Item
Overrange

Measurement characteristics

Resolution

Measurement frequency

Specification
Permits voltages that are up to 20 % over the range except
when the 750 Vrms range is in use
The maximum frequency for the 750 Vrms range is 100 kHz.

General specifications
Item
Specification
Input voltage range
100 Vac/120 Vac/220 Vac/240 Vac±10 %, single phase
Input frequency range
50 Hz/60 Hz ± 10 %Power consumption25 VAmax
Operating temperature range 0 °C to 50 °C
Operating humidity range
80 %rh or less (0 °C to 31 °C, no condensation)
Storage temperature range -40 °C to 70 °C (80 %rh or less, no condensation)
Operating altitude
Up to 2000 m
224(8.82)W×113(4.45)H×373(14.69)D mm(inch)
Dimensions/Weight
Approx. 3.7 kg (8.2 lb)
InterfaceUSB 2.0, GPIB (factory option)"Power cord" 1 pc. (with three-pronged plug), "Standard test leads"
Accessories(1 red, 1 black), "USB cable" 1pc., "Fuse"(spare) 1pc., "CD-ROM" *71pc., "Packing list,safety precautions" (1 English, 1 Japanese)
Complies with the requirements of the following directive andstandard.
Electromagnetic
EMC Directive 2014/30/EU
compatibility (EMC) *1, *2
EN 61326-1 (Class A *5)
EN 55011 (Class A *5, Group 1 *6)
EN 61000-3-2, EN 61000-3-3Complies with the requirements of the following directive andstandard.
Safety *1
Low Voltage Directive 2014/35/EU *2
EN 61010-1 (Class I *3, Pollution degree 2 *4), EN 61010-2-030*1 Does not apply to specially made or modified DME1600s. *2 Limited to products that have the CE/UKCA markon their panels. *3 This is a Class I equipment. Be sure to ground this product’s protective conductor terminal. The
safety of this product is only guaranteed when the product is properly grounded. *4 Pollution is addition of foreign
matter (solid, liquid or gaseous) that may produce a reduction of dielectric strength or surface resistivity. Pollution
Degree 2 assumes that only non-conductive pollution will occur except for an occasional temporary conductivity
caused by condensation. *5 This is a Class A equipment. This product is intended for use in an industrial environment. This product may cause interference if used in residential areas. Such use must be avoided unless the user
takes special measures to reduce electromagnetic emissions to prevent interference to the reception of radio andtelevision broadcasts. *6 This is a Group 1 equipment. This product does not generate and/or use intentionally radio-frequency energy, in the form of electromagnetic radiation, inductive and/or capacitive coupling, for the treatment of material or inspection/analysis purpose. *7 Contains the User’s Manual and the Remote Interface Manual.

■ Distributor:

1-1-3, Higashiyamata, Tsuzuki-ku, Yokohama, Kanagawa, 224-0023, Japan
Phone:(+81)45-593-0200, Facsimile:(+81)45-593-7591, https://global.kikusui.co.jp/KIKUSUI AMERICA, INC.
3625 Del Amo Blvd., Suite 160 Torrance, CA90503
Phone: 310-214-0000, Facsimile: 310-214-0014

KIKUSUI TRADING (SHANGHAI) Co., Ltd.
Room 305, Shenggao Building, No.137, Xianxia Road, Shanghai City, China
Phone: 021-5887-9067, Facsimile: 021-5887-9069

KIKUSUI ELECTRONICS EUROPE GmbH
Grossenbaumer Weg 8, 40472 Duesseldorf, Germany
Phone: +49(211)54257600, E-mail: support@kikusui-europe.com
For our local sales distributors and representatives, please refer to "sales network" of our website.

Issue:Jun.2024■ All products contained in this catalogue are equipment and devices that are premised on use under the supervision of qualifi ed
personnel, and are not designed or produced for home-use or use by general consumers. ■ Specifi cations, design and so forth
are subject to change without prior notice to improve the quality. ■ Product names and prices are subject to change and production
may be discontinued when necessary. ■ Product names, company names and brand names contained in this catalogue represent
the respective registered trade name or trade mark. ■ Colors, textures and so forth of photographs shown in this catalogue may
differ from actual products due to a limited fi delity in printing. ■ Although every effort has been made to provide the information as
accurate as possible for this catalogue, certain details have unavoidably been omitted due to limitations in space. ■ If you fi nd any
misprints or errors in this catalogue, it would be appreciated if you would inform us. ■ Please contact our distributors to confirm
specifications, price, accessories or anything that may be unclear when placing an order or concluding a purchasing agreement.

202406PDFEC51