Now eVerybody can build it...

Introducing eVance™ SPECT
Camera Building Blocks

Performance

The eVance™ Difference

SpatialThe medical community is adopting CZT-based detectors, andKromek is simplifying that adoption process with eVance™.

With a spatial resolution of 2.0mm andproprietary technology to subpixelize to
1.0mm resolution without compromising
efficiency, eVance™ is taking SPECT toa new level.
Use of CZT:n   Reduces patient dose

Plug & Play Technolog
We have made it simple todemonstrate, evaluate and integrate
eVance™ into your products.
Unlike other solutions, eVance™ is an
all-in-one package:n Improves sensitivity andspecificity

n Enables superior Multi-n CZT detectors
n Advanced ASICs
n Microelectronics and embedded

isotope SPECT

codeEnergy Resolution
The industry understands CZT offers
superior performance over standard
scintillator cameras. Kromek has taken
this to another level. Our eV-CZT™
and ASIC technology are providing
ground breaking energy resolution<4.5% FWHM at Tc-99 with excellent
efficiency.

n Packaged for scalabilityn Supports all camera architecture

eVance™ – TI
Thyroid Gamma Camera

eV-CZT™ Uniformity
One of the most distinct advantages of our technology is accuracy. eV has finetuned its brand of CZT crystals, enabling the most accurate detection results
available today.

140
Histogram of FWHM over eVance™
(4.4cm x 4.4cm, 484 pixels)

120
100

PIXELSeVance™ clearly out performs thetraditional Scintillator cameras.
8000

5mm thick THM CZT - single pixel - no corrections
57

7000

Co6000Counts

5000

8060

Pb Kα

4000

404.0%

3000

Pb Kβ

2000

20W Kα

1000
0020

4060

80

100Energy (keV)

Co57 Isotope measurement with Lead andTungsten radiation shields

1201400123456789FWHM [%]The above histogram illustrates the pixel to pixel resolution uniformity of an eVance™ module.

10CZT Building Blocks to Camera IntegrationeVance™ is a scalable solution for all SPECT applications including thyroid, breast, cardiac or
general purpose nuclear imaging.
n eVance™ – TI: Small form factor for positioning closer to the thyroidn eVance™ – BI: Able to see tumors near the chest wall for molecular breast imaging
n eVance™ – CI: Faster scanning and lower dose capabilities for cardiac imaging
n eVance™ – GP: Small, light weight package for general purpose SPECT

What sized camera do you need? eVance™ modules can be tiled into almost any size camera,
so you can tailor this technology to your future SPECT cameras.

eVance™ – TI camera head for Thyroid Imaging. (4x4 CZT Building Blocks.)

Over 10,000 eV-CZT™
detectors are sold annually into
the medical community
Kromek is the trusted partner in image detector technology, andwe will work closely with you to fit your imaging needs.

About Kromek Group plcFor over 25 years, Kromek has been a pioneer and leading manufacturer of CZT detector technology worldwide.
We serve many industries, including medical, homeland security and research. The entire Kromek team works toadvance CZT technology, and we support your business from engineering through implementation. Kromek has
made eV-CZT™ a high volume, commercially viable detector technology.

Kromek USA
373 Saxonburg Blvd.
Saxonburg, PA 16056
T: (724) 352-5288
E: sales@evproducts.com
EVC A4 SPF-DOC-1247 Rev1