Gaze Analyzing Perimeter

Gaze Analyzing Perimeter

MakingVisual Field Testing
Easier & More Accessible

Easy & Handy

Time-saving

Space-saving

This method reduces theexaminee's burden byeliminating the need for ﬁxation.

Quick and easy testing method
enables go-show/walk inexaminations.

The light-weight goggle-typemonitor eliminates the need fora darkroom and is easy to handle.

No central ﬁxation,
Reducing the burden on the examinee
L e ss St rain for Ex aminees
GAP-screener is a “gaze analysis type” of perimeter using eye-tracking technology,
a measurement method that allows examination simply by following the dotwith eyes. Since the system does not require long-term gaze fixation andbutton operation, it can prevent fatigue for the examinee.

L e ss W o rkload for Operat or s
Explain of the test procedure is easy. Since no button operatiosns by theexaminee, the operator does not need to stand by the examinee during thetest fot the entire time.

SimpleMaintenance
The head part can besanitized with alcohol

Gaze An a l y z i ng P e ri m e t e r :This system detects the center of the examinee's pupil in real time from images taken by an infrared camera
built into a head mounted monitor. This is a type of visual field perimeter that uses AI technology to analyze
the pupil's movements (gaze) as when they look at the dots displayed irregularly on the OLED display, thesystem determines whether the patient can see or not, based on saccadic and other features.

Space-saving, No darkroom required, Portable
Can be Used Anywhere
Only a PC and goggles are required to perform the test.
This dose not only save a lot of space as compared withconventional perimeters, it can be used anywhere, sincea dark room environment is not required.

In a comfortable position
A chinrest is not required, and the test can be performedin positions that were previously difficult for visual fieldtesting, such as bedsides, wheelchair, couches, and sofas.

Eye movement analysis and Response switch methods
Close to Objective Test
It is a near-objective test method analyzing eye movement
tracking in a reflective manner. This reduces the risk offalse positives, such as “I feel like I saw a target and pressed
a button.”

Corresponding to response switch
For examiners who are fatigued by eye movements orhave drooping eyelids, a test mode with central ﬁxation
and response switch is also available.

Time-based data reference
Equipped with a viewer
Since FINDEX has been providing ophthalmology solutions
such as the Claio image filing system, it can also provide

xx/xx/xx xxxxx
xx/xx/xx xxxxx
xx/xx/xx xxxxx

xxxx

xxxx

xxxx

system, the results can be viewed together with fundus
photographs and OCT.

xxxx

xxxx

xxxx
xxxx
xxxx

xxxx

xxxx

xxxx

xxxx

xxxx

xxxx

xxxx

xxxx

xxxxx
xxxxx
xxxx x x
xx xxx

xxxx

xxxx
xxxx
xxxx
xxxxxxxxxx
xxxxxxxxxxxxxxxxxx
xxxxxxxxxxxxxxxxxx
xxxxxxxxxxx
xxxxxxxxxxxxx
xx/xx/xx xxxxx
xxxxx
xxxxx

xxxxx

When used in combination with the Claio image filing

xxxx

xxxx

xxxxx

Scalability

xxxx

xx
xx

xx/xx/xx xxxxx

xxxx

xx
xx

and score graphs.

xx/xx/xx xxxxx

xxxx xxxx xxxx xxxx xxxx xxxx xxxx xxxx

xxxx

xx
xx
xx
xxxx
xx

allows users to view time series, comparative displays,xx/xx/xx xxxxx

xxxx

xx
xx

standard feature, we provide a dedicated viewer thatxx/xx/xx xxxxx

xx
xx

powerful support for referencing data over time. As axxxxx xxxxx xxxxx xxx
xxxx x xxxxx xx xxx
xx xx xx
xxx xxxx/xx/xx xxxxx

xxxxx
xxxxx

xx/xx/xx xxxxx

xxxxx

xxxxx

xxxxx

xxxxx

xxxxx

xxxxx

xxxxx

xxxxx

xxxxx

xxxxx

xxxxx

xx/xx/xx xxxxx
xx/xx/xx xxxxx
xx/xx/xx xxxxx
xx/xx/xx xxxxx
xx/xx/xx xxxxx
xx/xx/xx xxxxx
xx/xx/xx xxxxx
xx/xx/xx xxxxxxx/xx/xx xxxxx
xx/xx/xx xxxxx
xx/xx/xx xxxxx
xx/xx/xx xxxxx
xx/xx/xx xxxxx
xx/xx/xx xxxxx
xx/xx/xx xxxxx
xx/xx/xx xxxxx

xxxxx

xxxxx

xxxxx

xxxxx

xxxxx

Package contents

Reporting Images
Sensitivity measurement

1.25

0.25

Goggles

Correction Lens

Lends Holder

Storage Case
Screening

GAP
Threshold test
(Sensitivity measurement)

GAP-screener-

30 -2, 24-2,10 -2

Dark-spot detection test
(Screening)

30 -2, 24 - 2, 10 - 2

Size of visual indicator

GoldmanⅡ-Ⅴ

Maximum brightness

3,000cd/㎡

Body size (Goggle portion)

122(H)mm×201(W)mm×148(D)mm (Excluding Headband and Cables)

Body weight (Goggle portion)

400g (Excluding Cables)

Age of measurement
Details of corrective lenses

Age 13 and over
SPH±0.5〜10.0D
CYL±2.0〜6.0D

SPH±0.5〜10.0D

Product name: 視線分析型視野計 ゲイズ アナライジング ペリメーター GAP
Registration number : 38B２X10003000002
Product name: 視線分析型視野計 ゲイズ アナライジング ペリメーター GAP-screener
Registration number : 38B２X10003000003● Speciﬁcations and designs shown in this catalog are subject to change without prior notice.
● The contents of this catalog are as of June 2022.

FINDEX Inc.

https://findex.co.jp/en/

Tokyo Sankei Bldg 26F, 1- 7- 2 Otemachi, Chiyoda-Ku,
Tokyo, 100-0004 Japan
TEL : +81- 3 - 6271- 8958 FAX : +81- 3-6271- 8959