MultiSpect AnalysisTM
MultiSpect AnalysisTM

Gamma rayspectroscopy
softwareSpecialised gamma ray
spectroscopy software foranalysis of complex samples
for radionuclide identification
MultiSpect AnalysisTM has been designed specificallyfor Kromek’s range of radiation detectors on Windows
7 and 8 -based PC and tablets. It allows multiple
detectors to be connected and managed.
Acquisition from all detectors can be performed anddisplayed simultaneously, either viewing separate
spectra or combining the results together in a single
spectrum. Thumbnails are used to give an easy
visualisation of the signals from each detector. Spectra
can be displayed on an energy axis for fast and directcomparison of data.

Main featuresn

Identification of complex
samplesnMeasure the peak properties
and save data reports in .pdf
and .csv formatn

Clear colouring of the analysis
results for statistical significancenConnects and acquires
from multiple detectors
simultaneously

Multiple detectors and simultaneous acquisition
Analysis results:

Colour coded results showing which lines areabove the critical limit of the information.

n

Easily apply filters for viewing
the important datanConnects with GR1-A , GR1 ,SIGMATM and TN15TM via USBn

Library of gamma ray emissions
from over 400 nuclidesnSave spectra in the industry
standard .spe file formatnAdvanced radionuclide analysis
functionsnDirectly import measurementsfrom RayMon10TM

nInbuilt scripting allowing
automation with C# code

TMTM

Filter by critical limit, relative intensity of theemission lines, energy windows and half life
Included within MultiSpect AnalysisTM is a userconfigurable library of over 400 radionuclides. Resultsfrom matching spectra to the library are colour coded
according to statisticalsignificance for easy visualprocessing. The analysis
results can be filteredeasily and sorted to find therequired information.
Analysis of the spectrum
gives all the measured peak
information and can beexported in csv format for
use in other applications.

MultiSpect AnalysisTM
Specialised gamma ray spectroscopy
software for simultaneous acquisition
from multiple detectors
with advanced
radionuclide analysis
functionsApplications
FeatureHealth Physics

InSitu Monitoring

Nuclear Spectroscopy

Nuclear Industry

K-Spect

MultiSpect MultiSpect
AnalysisPremiumSpectral acquisition from single
GR1, GR1-A, SIGMA or TN15PPPSpectral acquisition frommultiple detectors of the same
type simultaneouslyOPP

Energy calibration facilityPPP

Efficiency calibration

OOP

Display multiple detector
information and more than 5saved SpectraOPP

Display calibrated spectra at
the same energy scales to allowcomparison

O

P

P

Compatible with K102
Multichannel Analyser

PPP

Thumbnail indication of loaded
spectraPPP

Ability to save spectra in SPE,CSV or N42.42 formats

PPP

Ability to export dataPPP

Ability to save detector
calibration informationPPP

Association of calibration datawith particular detectors byserial numberPPP

Aggregation of multiple spectra
into one spectrumOPP

Built in library of 416 isotopes

OPPIndustry standard categorisationof isotopes

OPP

Import from RayMon10OPP

Multiple regions of interest withSpectra

PP

P

Automated peak analysis ofSpectra

OPP

Quantitative analysis

OO

P

Drivers are available for both Windows (7 & 8) and Linux operating systems

MultiSpect Analysis Software
MultiSpect AnalysisTM software is written specificallyfor Kromek’s range of CZT-based semiconductor andscintillator radiation detectors for use with Linux orWindows 7 and 8 based PCs or tablets.

Kromek’s new generationof nuclear threat
detection
products...

MultiSpect AnalysisTM is Kromek’s feature-richsoftware. It receives the data and performs thespectrum acquisition, display, analysis and storage
functions.
Signals from the detectors are processed anddigitized, and the pulse height data is transferred tothe computer via USB.
MultiSpect AnalysisTM allows users to acquireand display live spectra from multiple devices
simultaneously alongside saved spectra from previous
measurements. It also enables grouping and summingof individual spectra plus the ability to match spectra
to an on-board library of over 400 nuclides.

SIGMA-50 (L) high sensitivity
scintillator detector andTN15 (R) high sensitivity non-Helium3
neutron detector

MultiSpect Analysis PremiumTM provides the added
benefit of efficiency calibration to enable quantitative
analysis ideally suited to data collected in any fixed
geometry such as marinelli beakers.

Analysis results:
Filter by critical limit, relative
intensity of the emission lines,
energy windows and half life.
Colour coded results showing
which lines are above a statisticalcritical limit.
A full export of the analysis results gives an easy to use.csv form of the analysis data with showing:n       Peak identificationnCentroidn

FWHM

n

FWnM at user configurable leveln

FW rationGross Countsn

Net CountsnStapleton Critical LimitnUpper and lower confidence limitsnEquations and parameters of thesubtracted backgroundn

Clear details of the peak andbackground region edge locations.

Configuration options for the analysis
based on statistical significance.

User configurable library ofradionuclide emissions:T: +44 (0) 1740 626060 F: +44 (0) 1740 626061 E: sales@kromek.com

W: www.kromek.com

MSA A4E SPF-DOC-1221 Rev8Kromek Ltd NetPark Thomas Wright Way Sedgefield County Durham TS21 3FD UK