TD109-0

GAS DETECTION
HAND BOOK〒174-8744 東京都板橋区小豆沢 2-7-6
ホームページ http://www.rikenkeiki.co.jp/

Contents
1. はじめに(Introduction) .............................................................................................................................. 4
2. ガスとは何か?(What is gas?) ...................................................................................................................... 5
2-1. 大気の成分(Air composition) ............................................................................................................... 5
2-2. ガスの危険性(Gas hazards) ................................................................................................................... 6
3. 可燃性ガスの危険性（Flammable Gas hazards） .................................................................................. 7
3-1. 燃焼の三要素(Flammable gase date) ................................................................................................ 7
3-2. 爆発限界(Explosive Limit) ................................................................................................................... 8
3-3. 可燃性蒸気 ............................................................................................................................................. 8
3-4. 可燃性ガスの物性(Flammable gase date) ......................................................................................... 10
4. 毒性ガスの危険性(Toxic Gas hazards) ................................................................................................... 29
4-1. 毒性ガスの危険性 .................................................................................................................................. 29
4-2. 許容濃度(Toxic Gas) ........................................................................................................................... 30
5. 酸素欠乏の危険性（Oxygen deficiency） ............................................................................................. 39
5-1. 酸素欠乏............................................................................................................................................... 39
5-2. 酸素欠乏の主な 3 つの要因.................................................................................................................... 40
5-3. 過剰酸素............................................................................................................................................... 40
6. ガス検知器の用途・設置場所(Typical areas that require gas detection) ........................................ 41
6-1. ガス検知器の市場 .................................................................................................................................. 41
7. ガス検知技術(Gas detection technologies) ........................................................................................ 45
7-1. 理研計器のガスセンサ技術 ...................................................................................................................... 45
7-2. 接触燃焼式(Catalytic Combustion Method) .................................................................................... 46
7-3. ニューセラミック式(Catalytic Combustion Method) .............................................................................. 47
7-4. 半導体式(Semi-Conductor Method) ................................................................................................ 48
7-5. 熱線型半導体式(Hot Wire Type Semi-Conductor Method ).......................................................... 49
7-6. 熱伝導式(Thermal Conductivity Method) ....................................................................................... 50
7-7. 定電位電解式(Potentiostatic Electrolysis Method)........................................................................ 51
7-8. 隔膜分離型定電位電解式(Membrane-Separated Electrode Method ) .......................................... 52
7-9. 隔膜ガルバニ電池式(Membrane Type Galvanic Cell Method ) ........................................................ 53
7-10. 非分散型赤外線式(Non-Dispersive Infrared Method) .................................................................. 54
7-11. 光波干渉式(Non-Dispersive Infrared Method)............................................................................. 55
7-12. 検知テープ式(Chemical Tape Method) ............................................................................................ 56
7-13.光イオン化式センサ(Photo-Ionization Detector) ................................................................................ 57
7-14.熱粒子化式センサ(Pyrolysis-Particle Detection Method) ................................................................ 58
8. ガス検知器の選定(Selecting the proper measurement method) .................................................. 59
8-1. ガス検知器の選定 .................................................................................................................................. 59
9. 定置式ガス検知器(Fixed Gas Detectors) ............................................................................................... 61
9-1. 防爆型ガス検知器.................................................................................................................................. 61
9-2. 非防爆型ガス検知器 .............................................................................................................................. 68
9-3. 指示警報部 ........................................................................................................................................... 73
10. ポータブルガス検知器(Portable Gas detectors) .................................................................................. 75
10-1. 複合型ガス検知器(Multi-sensor) ...................................................................................................... 75-2-

10-2. 単成分ガス検知器(Single-sensor) .................................................................................................... 82
11. 代表的なアプリケーション(Typical applications for portable gas detectors) ................................ 87
11-1. 自動車市場のアプリケーション ................................................................................................................. 87
11-3. 鉄鋼市場のアプリケーション .................................................................................................................... 94
11-4. 燻蒸市場のアプリケーション .................................................................................................................... 98
11-5. 製薬市場のアプリケーション .................................................................................................................. 101
11-6. 食品市場のアプリケーション .................................................................................................................. 106
11-7. 大学・研究施設のアプリケーション ......................................................................................................... 114
11-8. ガス市場のアプリケーション .................................................................................................................... 120
11-9. 電力市場のアプリケーション .................................................................................................................. 124
11-10. 製紙市場のアプリケーション ................................................................................................................ 134
11-11. 石油・石油化学市場のアプリケーション ................................................................................................ 139
11-12. 上下水道市場のアプリケーション ........................................................................................................ 145
11-13. 電子・デバイス市場のアプリケーション .................................................................................................. 153
12. 国際規格および認証（Global standards and approvals） ............................................................ 161
12-1. IEC 規格 ........................................................................................................................................... 161
12-2. IECEx ............................................................................................................................................... 161
12-3. 理研計器製品に関わる IEC 規格 ....................................................................................................... 162
12-4. ISO 規格 ........................................................................................................................................... 163
12-5. 理研計器製品に関わる ISO 規格 ....................................................................................................... 163
13. 地域規格(Regional standards) ......................................................................................................... 164
13-1. 欧州指令と CE マーキング ................................................................................................................... 164
13-2. 理研製品に関わる欧州指令 ............................................................................................................... 165
13-3. 欧州規格(EN 規格) .......................................................................................................................... 165
13-4. 理研計器製品に関する EN 規格......................................................................................................... 166
13. 国家規格(National standards) ......................................................................................................... 168
14-１. 日本 ................................................................................................................................................ 168
14-2. 理研計器製品に関する EN 規格......................................................................................................... 170
14-3. 韓国―KCs マーク .............................................................................................................................. 171
14-4. ブラジル―INMETRO 認証 ................................................................................................................. 171
14-5. アメリカ ............................................................................................................................................... 171
13. 機器の保護等級(Ingress protection of enclosures)..................................................................... 172
14. メンテナンスと保守について（Gas detection maintenance and ongoing care） .......................... 174-3-1. はじめに(Introduction)
2-1. 大気の成分(Air composition)1
1.

はじめに(Introduction)

RIKENKEIKI GAS HAND BOOK は、定置型や可搬型のガス検知器の使用を検討しているお客様に対して、ガス検知を単純かつ包括的に理解していただくための、リファレンスガイドの目的で作成しております。
理研計器のガス検知警報器は、災害を引き起こす可燃性ガス、毒性ガス、酸素欠乏をいち早く検知し、警報を発することで、目に見えない危険から人々を守っています。
ガス検知器は大きく分けると固定した場所で使用する定置式と、携帯して持ち運びができるポータ
ブル型があり、ガスを使用する幅広い産業の用途・ニーズに合わせて、様々な種類のガス検知器やガスセンサをラインナップしております。
本書ではガスの種類や危険性、ガス検知器やガスセンサの種類や特性の詳細について説明すると
ともに、産業毎のアプリケーション例をご紹介しています。本書をお読みいただくことで、ガス検知器をより深く理解していただき、お客様のガス検知器の選定や、ガスによる災害を未然に防ぐために、ガス検知器を最大限に活用することにお役立ていただければ幸いです。
私たちの究極の使命は、世の中からガス事故をなくすことです。"人々が安全に働ける環境をつくる"という揺るぎない理念のもと、新しい技術で産業を支え、新たな挑戦を続けてまいります。

-4-

2. ガスとは何か?(What is gas?)2-1. 大気の成分(Air composition)22.

ガスとは何か?(What is gas?)2-1. 大気の成分(Air composition)
空気の構成は窒素が約 78％、酸素が約 21％、その他（アルゴン、二酸化炭素など）が約１％です。最も多くを占める窒素は、アミノ酸をはじめとする多くの生体物質の中に含まれたんぱく質の素となり、地球のほぼ全ての生物にとって必要な働きをしています。しかし窒素は空気中から直接からだに取り込まれることはなく、呼吸によってとりいれられた窒素は、また呼吸によって体外へ排出されます。私たちのからだに直接取り込まれ生命維持に欠かせない働きをする酸素は 21％であり、植物の光合成に必要な二酸化炭素はわずか 0.04％に過ぎません。動物は酸素を取り入れ二酸化炭素を排出し、植物は二酸化炭素を取り入れ酸素を排出して、空気全体のバランスを保ちながら生命を営み維持しています。
大気の主要成分成分

化学式

体積割合[%]

質量割合[%]

窒素

N2

78.08475.524酸素

O2

20.9476

23.139

アルゴン

Ar

0.934

1.288

二酸化炭素

CO2

0.03140.0477ネオン

Ne0.001818

0.00127

ヘリウム

He

0.000524

0.000072

クリプトン

Kr

0.000114

0.00033

キセノン

Xe

0.0000087

0.000039

水素H20.00005

0.000003

メタン

CH40.00020.0001

一酸化二窒素N2O

0.00005

0.00008

ISO 2533-1975 より-5-

2. ガスとは何か?(What is gas?)2-2. ガスの危険性(Gas hazards)

2-2. ガスの危険性(Gas hazards)
ガスの危険性には主に下記 3 つがあります。
①可燃性ガス
20℃、標準気圧 101.3 kPa において、空気と混合した場合に爆発範囲（燃焼範囲）を もつガス
*GHS（Globally Harmonized System of Classificationand Labelling of Chemicals)の定義よる②毒性ガス
人間が吸引したり、触れたりすると人体に害を及ぼすガス
労働現場においては、労働者が一日 8 時間、
週 40 時間暴露した場合に健康上の被害を及ぼす値を閾値として、毒性ガスを管理する。

②酸素欠乏
人間の体は通常、大気中の酸素約 21％中で正常に活動できるよう出来ています。酸素の消費(金属の酸化や微生物)や他ガス(N2、Ar)による酸素の置換が起こり、酸素濃度が低下した場合、約 18％を下回る濃度から人体に支障を来し始め、6～8％になると命に関わる状態になります。

-6-

3. 可燃性ガスの危険性（Flammable Gas hazards）
3-1. 燃焼の三要素(Flammable gase date)33.

可燃性ガスの危険性（Flammable Gas hazards）
3-1. 燃焼の三要素(Flammable gase date)
一般的に燃焼とは熱と光を伴う酸化反応(対象物質が酸素と結合する)のことをいいます。
物資が燃焼するためには①可燃性物質②支燃性ガス③着火源が必要であり、どれか一つでも欠ければ燃焼しません。そのため、ガスを燃焼させないように考えた場合、ガスが燃える濃度：支燃性ガ
スと着火源があった場合に燃焼しうる濃度にならないように管理することが最も重要になります。

-7-

3. 可燃性ガスの危険性（Flammable Gas hazards）
3-2. 爆発限界(Explosive Limit)

3-2. 爆発限界(Explosive Limit)
可燃性ガスまたは可燃性液体の蒸気が空気または酸素と混合した場合，特定のガス濃度範囲
で着火源が存在するとき爆発する。この濃度範囲を爆発範囲といい，濃度の低いほうの限界を爆発
下 限 界 (LEL ： Lower Explosion Limit) ， 高 い ほ う の 限 界 を 爆 発 上 限 界 (UEL: Upper
Explosion Limit)という。
Ex. 水素の場合

爆発下限界値は実証試験によって定められる値ですが、試験条件・方法によって結果は変わることがあります。そのため、参照する文献によって値が異なる場合がありますので注意が必要です。
可燃性ガスが爆発上限界を超える高濃度だとしても、大気中に漏洩した場合に直ちに薄まりながら拡散し、爆発範囲内の危険な濃度になることから、一般的なガス検知器では爆発下限界でガス濃度を管理します。爆発下限界値を 100 とした場合の割合である％LEL という単位が用いられます。

3-3. 可燃性蒸気
一般的には同じ気体でもガスと蒸気といった場合、その意味合いは異なる。蒸気といった場合、常温では液体（または個体）で存在する物質がある条件の下で液相から気層へ気化した状態のことを指す。ここで可燃性蒸気が危険な状態になりうるかどうかは温度変化に基づく下記の物性値がその指標となる。
1. 飽和蒸気圧
ある一定温度で物質が液体から気体に気化するときの圧力のこと。一般的に温度が上昇するほど蒸気圧は大きい値になる。また、大気圧(101.3ｋPa≒760 ㎜ Hg)と同じ圧力になる温度を沸点といい、大気圧に対する蒸気圧の 100 分率を計算すると、その温度でどの位のガス濃度(体積濃度)が気化し得るかを知ることができる。
次頁に例としてエタノールと水の飽和蒸気圧曲線を示す。水の沸点は 100℃であるから、蒸気圧曲線が 101.3kPa のとき 100℃を示していることがわかるだろう。つまり、このときの飽和水蒸気濃度は-8-

3. 可燃性ガスの危険性（Flammable Gas hazards）
3-3. 可燃性蒸気

100vol％ということになる。一方でエタノールは、病院で注射を打つときの脱脂綿に含ませる消毒エタノールと水の飽和蒸気圧曲線用のエタノールを想像すれば容易に想像できるが、水よりも気化しやすい(≒蒸気圧の高い)液体

エタノールである。実際にエタノールの沸点は約 78℃であ水り、データからも水よりも気化しやすいことがわかる。
ここで、ある温度でのエタノールのガス濃度を知りたい場合、その温度での蒸気圧から算出できる。
例えば、飽和蒸気圧曲線から 20℃のときの蒸気圧は約 5.8kPa と読み取れる。それを以下の計算式によってガス濃度を導き出せる。

ガス濃度(体積％濃度) ＝ [ある温度で蒸気圧] ÷ (大気圧) × 100＝ 5.8(kPa) ÷ 101.3(kPa) × 100≒ 5.7 vol％この計算は仮に上記のような蒸気圧曲線がなくても薬品メーカーの SDS(安全データシート)には一般的に蒸気圧データが常温(20℃～30℃)等のポイントで記載されており、その値からガス濃度を算出できるので覚えておくと便利である。

2. 引火点
引火点とは物質が気化して空気と混合した場合に引火する濃度になる最低温度のことをいう。言い換え
れば、可燃性蒸気が LEL 濃度に達するときの温度とも解釈できる。可燃性蒸気を発生する液体の引火危険性を考えた場合に、使用雰囲気温度よりも低い場合は燃焼または爆発事故の可能性が高いので注意が必要である。

3. 発火点
空気中の可燃物が一般に着火源と呼ばれる電気火花や火炎、あるいは赤熱金属線のような局部的な高温物との接触による発火の場合ではなく、その物質自身の温度上昇によって自然に発火する場合の最低温度のことをいう。防爆電気機器のメーカーは可燃性ガス・蒸気に触れる恐れのある電気機器等の表面温度が、対象となるガス・蒸気の発火温度を超えないように設計・製作する必要がある。

-9-

3. 可燃性ガスの危険性（Flammable Gas hazards）
3-4. 可燃性ガスの物性(Flammable gase date)

3-4. 可燃性ガスの物性(Flammable gase date)
下表に、可燃性ガスの爆発下限界値を①ISO 10156(2010) ②IEC60079-20-1(2020)②理研計器標準値の 3 種類を示します。
ISO
10156 4th ed.

IEC60079-20-1
LEL(vol%)

理研計器
標準

-

7.0

7.0

-

1.61

-

2.4

1,1´-Oxybisethane(CH3CH2)2O

1.7

1.7

1.7

ｱﾆﾘﾝ

Benzenamine
C6H5NH2

-

1.2

1.3

64-17-5ｴﾁﾙｱﾙｺｰﾙ

Ethanol
CH3CH2OH

3.1

3.1

3.3

7

64-18-6ｷﾞ酸

Formic Acid
HCOOH

-

18.0

18.0

8

64-19-7酢酸

Acetic acid
CH3COOH

-

4.0

4.0

9

64-67-5硫酸ｼﾞｴﾁﾙ

Sulfuric acid diethyl ester(CH3CH2)2SO4

-

-

10

67-56-1ﾒﾁﾙｱﾙｺｰﾙ

Methanol
CH3OH

6.0

6.0

5.5

11

67-63-0

ｲｿﾌﾟﾛﾋﾟﾙｱﾙｺｰﾙ

2-Propanol(CH3)2CHOH

-

2.0

2.0

12

67-64-1ｱｾﾄﾝ

2-Propanone(CH3)2CO

2.5

2.5

2.15

13

68-12-2

N,N-ｼﾞﾒﾁﾙﾎﾙﾑｱ
ﾐﾄﾞN,N-Dimethyl formamide
HCON(CH3)2

-

1.8

14

71-23-8

ﾉﾙﾏﾙﾌﾟﾛﾋﾟﾙｱﾙ
ｺｰﾙ

1-Propanol
CH3CH2CH2OH

-

2.1

2.1

15

71-36-3

ﾉﾙﾏﾙﾌﾞﾁﾙｱﾙｺｰ
ﾙ1-Butanol
CH3(CH2)2CH2OH

-

1.4

1.4

16

71-41-0

ｱﾐﾙｱﾙｺｰﾙ

1-Pentanol
CH3(CH2)3CH2OH

-

1.06

1.2

17

71-43-2ﾍﾞﾝｾﾞﾝ

Benzene
C6H6

1.2

1.2

1.2

18

74-82-8ﾒﾀﾝ

Methane
CH4

4.4

4.4

5.0

19

74-84-0ｴﾀﾝ

EthaneCH3CH3

2.4

2.4

3.0

20

74-85-1ｴﾁﾚﾝ

Ethene(CH2=CH2

2.4

2.3

2.7

21

74-86-2ｱｾﾁﾚﾝ

Ethine
CH≡CH

2.3

2.3

1.5

22

74-87-3

塩化ﾒﾁﾙ

Methyl chloride
CH3Cl

7.6

7.6

8.1

74-89-5ﾓﾉﾒﾁﾙｱﾐﾝ

Methylamine
CH3NH2

4.9

4.2

4.9

74-90-8

ｼｱﾝ化水素Hydrocyanic acid
HCN

5.4

5.4

5.6

NO.

CAS-No.

ｶﾞｽ名

Name formula

150-00-0

ﾎﾙﾏﾘﾝ
ﾎﾙﾑｱﾙﾃﾞﾋﾄﾞ251-80-9

N,N,N‘,N’－ﾃﾄﾗ
ﾒﾁﾙｼﾞｱﾐﾉﾒﾀﾝ

357-14-7

1,1-ｼﾞﾒﾁﾙﾋﾄﾞﾗ
ｼﾞﾝ

Formaldehyde
HCHO
N,N,N',N'-Tetramethyl
methanediamine(CH3)2NCH2N(CH3)2
1,1-Dimethylhydrazine(CH3)2NNH2

460-29-7ｼﾞｴﾁﾙｴｰﾃﾙ

562-53-3

6

23
24- 10 -

3. 可燃性ガスの危険性（Flammable Gas hazards）
3-4. 可燃性ガスの物性(Flammable gase date)NO.

CAS-No.

ｶﾞｽ名

Name formulaISO
10156 4th ed.

IEC60079-20-1
LEL(vol%)

理研計器
標準

25

74-93-1ﾒﾁﾙﾒﾙｶﾌﾟﾀﾝ

Methanethiol
CH3SH

4.1

4.1

3.9

26

74-96-4

臭化ｴﾁﾙ

Bromoethane
CH3CH2Br

-

6.7

27

74-98-6

ﾌﾟﾛﾊﾟﾝ

Propane
CH3CH2CH3

1.7

1.7

28

74-99-7

ﾌﾟﾛﾋﾟﾝ

Propyne
CH3C≡CH

1.8

1.7

29

75-00-3

ﾓﾉｸﾛﾛｴﾀﾝ

Chloroethane
CH3CH2Cl

3.6

3.6

3.6

30

75-01-4

塩化ﾋﾞﾆﾙ

Chloroethene
CH2=CHCl

3.8

3.6

3.8

31

75-04-7

ﾓﾉｴﾁﾙｱﾐﾝ

Ethylamine
C2H5NH2

3.5

3.5

3.5

32

75-05-8

ｱｾﾄﾆﾄﾘﾙ

Acetonitrile
CH3CN

-

3.0

3.0

33

75-07-0

ｱｾﾄｱﾙﾃﾞﾋﾄﾞ

Ethanal
CH3CHO

4.0

4.0

4.0

34

75-08-1

ｴﾁﾙﾒﾙｶﾌﾟﾀﾝ

Ethanethiol
CH3CH2SH

-

2.8

2.8

35

75-15-0

二硫化炭素

Carbon Disulfide
CS2

0.6

0.6

1.3

36

75-19-4ｼｸﾛﾌﾟﾛﾊﾟﾝ

Cyclopropane
CH2CH2CH2

2.4

2.4

37

75-21-8

酸化ｴﾁﾚﾝ

Oxirane
CH2CH2O

2.6

2.6

3.0

38

75-28-5ｲｿﾌﾞﾀﾝ

2-Methylpropane(CH3)2CHCH3

1.5

1.3

1.8

39

75-29-6ｲｿﾌﾟﾛﾋﾟﾙｸﾛﾗｲﾄﾞ

2-Chloropropane(CH3)2CHCl

-

2.8

2.8

40

75-31-0

ｲｿﾌﾟﾛﾋﾟﾙｱﾐﾝ

2-Propaneamine(CH3)2CHNH2

-

2.3

41

75-34-3

1,1-ｼﾞｸﾛﾛｴﾀﾝ

-

5.6

42

75-35-4

1,1-ｼﾞｸﾛﾛｴﾁﾚﾝ

-

6.5

43

75-36-5

塩化ｱｾﾁﾙ-

5.0

75-38-7

ﾌｯ化ﾋﾞﾆﾘﾃﾞﾝﾓﾉ
ﾏｰ

1,1-Difluoroethene
CH2=CF2

4.7

3.9

2.3

45

75-50-3ﾄﾘﾒﾁﾙｱﾐﾝ

Trimethylamine(CH3)3N

2.0

2.0

2.0

46

75-52-5

ﾆﾄﾛﾒﾀﾝ

Nitromethane
CH3NO2-

7.3

4775-56-9酸化ﾌﾟﾛﾋﾟﾚﾝ

1.9

1.9

75-83-2ｼﾞﾒﾁﾙﾌﾞﾀﾝ

1.2

1.0

49

75-85-42-ﾒﾁﾙ-2-ﾌﾞﾀﾉｰﾙ

2-Methyloxirane
CH3CHCH2O
2,2-Dimethylbutane(= Neohexan)(CH3)3CCH2CH3
2-Methylbutan-2-ol
CH3CH2C(OH)(CH3)2

-

1.4

50

75-86-5ｱｾﾄﾝｼｱﾝﾋﾄﾞﾘﾝ

2-Hydroxy-2-methyl-propionitrile
CH3 C(OH)CNCH3

-

2.2

2.2

51

75-89-8

ﾄﾘﾌﾙｵﾛｴﾀﾉｰﾙ

2,2,2-Trifluoroethanol
CF3CH2OH

-

8.4

5.5

52

76-37-9ﾌｯ素ｱﾙｺｰﾙ N-1

2,2,3,3-Tetrafluoropropan-1-ol
HCF2CF2CH2OH

-

-

4.8

44

48

1,1-Dichloroethane
CH3CHCl2
1,1-Dichloroethene(= Vinylidene Chloride)
CH2=CCl2
Acetyl chloride
CH3COCl- 11 -

2.0

5.6

2.8

3. 可燃性ガスの危険性（Flammable Gas hazards）
3-4. 可燃性ガスの物性(Flammable gase date)NO.

CAS-No.

ｶﾞｽ名

5377-73-6ｼﾞｼｸﾛﾍﾟﾝﾀｼﾞｴﾝ

54

77-78-1

硫酸ｼﾞﾒﾁﾙ

55

78-10-4ﾃﾄﾗｴﾄｷｼｼﾗﾝ

56

78-78-4ｲｿﾍﾟﾝﾀﾝ57

78-80-8ｲｿﾌﾟﾛﾍﾟﾆﾙｱｾﾁﾚ
ﾝ

58

78-81-9ｲｿﾌﾞﾁﾙｱﾐﾝ

5978-83-1

ｲｿﾌﾞﾁﾙｱﾙｺｰﾙ60

78-84-2ｲｿﾌﾞﾁﾙｱﾙﾃﾞﾋﾄﾞ61

78-86-4

2-ｸﾛﾛﾌﾞﾀﾝ

6278-87-5

1,2-ｼﾞｸﾛﾛﾌﾟﾛﾊﾟﾝ

63

78-92-2

2-ﾌﾞﾁﾙｱﾙｺｰﾙ64

78-93-3ﾒﾁﾙｴﾁﾙｹﾄﾝ

65

79-09-4

ﾌﾟﾛﾋﾟｵﾝ酸

6679-10-7

ｱｸﾘﾙ酸

67

79-20-9

酢酸ﾒﾁﾙ

Name formula
3a,4,7,7a-Tetrahydro-4,7-methano1H- indene(= Dicyclopentadiene)(= Cyclopentadiene dimer)
C10H12
Sulfuric acid dimethyl ester(= Dimethyl sulfate)(CH3O)2SO2
Tetraethoxy Silane(= Silicic acid tetraethyl ester)(= Tetraethyl silicate)(= Silicon tetraethoxide)(C2H5)4Si
2-Methylbutane(= Ethyl dimethyl methane)(= Isopentane)(CH3)2CHCH2CH3
2-Methyl-1-buten-3-yne
HC≡CC(CH3)CH2
2-Methylpropan-1-amine(= iso-Butylamine)(CH3)2CHCH2NH2
2-Methyl-1-propanol(= iso-Butanol)(= iso-Propylcarbinol)(= iso-Butyl alcohol)(CH3)2CHCH2OH
2-Methyl-1-propanal(= iso-Butanal)(= iso-Butyraldehyde)(CH3)2CHCHO
2-Chlorobutane(= sec-Butyl chloride)
CH3CHClCH2CH3
1,2-Dichloropropane(= Propylene dichloride)
CH3CHClCH2Cl
2-Butanol(= sec-Butyl alcohol)(= Butylene hydrate)(= 2-Hydroxybutane)(= Methyl ethyl carbinol)
CH3 CHOHCH2 CH3
2-Butanone(= Ethyl methyl ketone)(= Methyl acetone)(= Methyl ethyl ketone)
CH3CH2COCH3
Propionic acid(= Carboxyethane)(= Ethanecarboxylic acid)(= Methyl acetic acid)
CH3CH2COOH
2-Propenoic acid(= Acroleic acid)(= Ethylenecarboxylic acid)(= Glacial acrylic acid)(= Acrylic acid)
CH2=CHCOOH
Acetic acid methyl ester(= Methyl acetate)(= Ethanoic acid methyl ester)(= Methyl ethanoate)
CH3COOCH3- 12 -

ISO
10156 4th ed.

IEC60079-20-1
LEL(vol%)

理研計器
標準

-

0.8

1.0

-

-

-

0.45

1.4

1.3

1.3

1.3

-

1.4

-

1.47

-

1.4

1.7

-

1.6

1.6

-

2.0

-

3.4

3.4

-

1.7

1.7

1.5

1.5

1.8

-

2.1

-

2.4

3.0

3.1

3.1

3.1

3. 可燃性ガスの危険性（Flammable Gas hazards）
3-4. 可燃性ガスの物性(Flammable gase date)NO.

CAS-No.

ｶﾞｽ名

6879-22-1ｸﾛﾛｷﾞ酸ﾒﾁﾙ

6979-24-3

ﾆﾄﾛｴﾀﾝ70

79-29-8

2,3-ｼﾞﾒﾁﾙﾌﾞﾀﾝ

71

79-31-2ｲｿ酪酸

7279-38-9ｸﾛﾛﾄﾘﾌﾙｵﾛｴﾁﾚﾝ

73

80-62-6

ﾒﾀｸﾘﾙ酸ﾒﾁﾙ

7491-20-3

ﾅﾌﾀﾚﾝ

75

95-47-6

ｷｼﾚﾝ

76

95-92-1

ｼｭｳ酸ｼﾞｴﾁﾙ

77

96-22-0ｼﾞｴﾁﾙｹﾄﾝ

78

96-33-3ｱｸﾘﾙ酸ﾒﾁﾙ7996-37-7ﾒﾁﾙｼｸﾛﾍﾟﾝﾀﾝ

80

97-62-1ｲｿ酪酸ｴﾁﾙ

81

97-63-2

ﾒﾀｸﾘﾙ酸ｴﾁﾙ

8297-85-8ｲｿ酪酸ｲｿﾌﾞﾁﾙ

83

97-88-1

ﾒﾀｸﾘﾙ酸ﾉﾙﾏﾙﾌﾞ
ﾁﾙ

Name formula
Carbonochloridic acid methyl ester(= Methyl chloroformate)(= Methoxycarbonyl chloride)
CH3OOCCl
Nitroethane
CH3CH2NO2
2,3-Dimethylbutane(= Diisopropyl)(CH3)2CH(CH3)CH2CH3
2-Methylpropanoc acid(= iso-Butyric acid)(= Dimethylacetic acid)(CH3)2CHCOOH
Chlorotrifluoroethene(= Chlorotrifluoroethylene)
CF2=CFCl
2-Methyl-2-propenoic acid methyl ester(= Methyl methacrylate)(= Methacrylate monomer)(= Methyl ester of methacrylic acid)(= Methyl-2-methyl-2-propenoate)
CH3=CCH3COOCH3
Naphthalene(= Tar camphor)(= White tar)
C10H8
1,2-Dimethyl benzene(= o-Xylene)(= o-Xyol)
C6H4(CH3)2
Ethanedioic acid diethyl ester(= Diethyl Oxalate)(= Oxalic acid diethyl ester)(COOCH2CH3)2
Pentan-3-one(= Diethyl ketone)(= Metacetone)(= Propione)(CH3CH2)2CO
Propenoic acid methyl ester(= Acrylic acid methyl ester)(= Methoxycarbonyl ethylene)(= Methyl propenoate)(= Methyl Acrylate)
CH2=CHCOOCH3
Methylcyclopentane
CH3CH(CH2)3CH2.
2-Methylpropanoic acid ethyl ester(= Ethyl isobutyrate)(= Ethyl 2-methylpropanoate)(CH3)2CHCOOC2H5
2-Methyl-prop-2-enoic acid ethyl ester(= Methacrylic acid ethyl ester)(= Ethyl methacrylate)
CH2=CCH3COOCH2CH3
2-Methylpropanoic acid 2-methylpropyl
ester(= iso-Butyl isobutyrate)(CH3)2CHCOOCH2CH(CH3)2
2-Methyl-2-propenoic acid butyl ester(= Butyl methacrylate)(= Butyl-2-methylprop-2-enoate)
CH2=C(CH3)COO(CH2)3CH3- 13 -

ISO
10156 4th ed.

IEC60079-20-1
LEL(vol%)

理研計器
標準

-

7.5

7.5

-

3.4

-

1.0

-

2.0

4.6

4.6

8.4

-

1.7

1.7

-

0.6 at 150°C

-

1.0

-

-

-

1.6

1.6

-

1.95

2.4

-

1.0

-

1.6

-

1.5

-

0.8

-

1.0

1.0

2.0

3. 可燃性ガスの危険性（Flammable Gas hazards）
3-4. 可燃性ガスの物性(Flammable gase date)NO.

CAS-No.

ｶﾞｽ名

8497-95-02-ｴﾁﾙﾌﾞﾀﾉｰﾙ

85

97-99-4ﾃﾄﾗﾋﾄﾞﾛﾌﾙﾌﾘﾙｱ
ﾙｺｰﾙ

86

98-00-0

ﾌﾙﾌﾘﾙｱﾙｺｰﾙ87

98-01-1

ﾌﾙﾌﾗｰﾙ88

98-82-8

ｸﾒﾝ

89

98-83-9α-ﾒﾁﾙｽﾁﾚﾝ

9098-95-3

ﾆﾄﾛﾍﾞﾝｾﾞﾝ

91

99-87-6

p-ｼﾒﾝ

92

100-37-8

2-ｼﾞｴﾁﾙｱﾐﾉｴ ﾀ
ﾉｰﾙ

93

100-40-3

4-ﾋﾞﾆﾙｼｸﾛﾍｷｾﾝ

94

100-41-4ｴﾁﾙﾍﾞﾝｾﾞﾝ

95

100-42-5ｽﾁﾚﾝ

96100-43-6

4-ﾋﾞﾆﾙﾋﾟﾘｼﾞﾝ

97100-44-7塩化ﾍﾞﾝｼﾞﾙ

98

100-52-7

ﾍﾞﾝｽﾞｱﾙﾃﾞﾋﾄﾞ

Name formula
2-Ethyl-1-butanol(= Isohexyl alcohol)
CH3CH(CH2CH3)CH2CH2OH
Tetrahydro-2-furan methanol)(= Tetrahydrofurfuryl alcohol)(= Tetrahydrofuran-2-yl-methanol)(= Tetrahydro-2-furan carbinol)(= 2-Hydroxymethyl oxolane)
OCH2CH2CH2CHCH2OH
2-Furylmethanol(= Furfuryl Alcohol)(= 2-Hydroxymethylfuran)
OC(CH2OH)CHCHCH
2-Furancarbox aldehyde(= Fural)(= Furfural)(= 2-Furaldehyde)
OCH=CHCH=CHCHO(1-Methylethyl) benzene(= Cumene)(= Isopropyl benzene)(= 2-Phenyl propane)
C6H5CH (CH3)2
α-Methyl styrene(= Isopropenyl benzene)(= 1-Methyl-1-phenylethylene)(= 2-Phenyl propylene)
C6H5C(CH3)=CH2
Nitrobenzene(= Nitrobenzol)(= Oil of mirbane)
C6H5NO2
1-Methyl-4-(1-methylethyl)benzene(= p-Cymene)(= p-isopropyltoluene)
CH3C6H4CH (CH3)2
2-Diethylaminoethanol(= Diethylaminoethanol)(= 2-Diethylaminoethyl alcohol)(= N,N-Diethylethanol amine)(= Diethyl-(2-hydroxyethyl)amine)(= 2-Hydroxytriethylamine)(C2H5)2NCH2CH2OH
4-Ethenylcyclohexene(= Vinyl cyclohexene)(CH2=CH)CH(CH2)4CH2
Ethylbenzene(= α-Methyltoluene)(= Phenylethane)
C6H5CH2CH3
Ethenylbenzene(= Styrene)(= Vinylbenzene)(= Phenylethylene)(= Styrol)
C6H5CH=CH2
4-Vinylpyridine(= 4-Ethenylpyridine)(= γ-Vinylpyridine)
NCHCHC(CH2=CH)CHCH(Chloromethyl)benzene (= Benzyl
chloride)(= α-Chlorotoluene)(= Tolyl chloride) C6H5CH2Cl
Benzaldehyde
C6H5CHO- 14 -

理研計器
標準

ISO
10156 4th ed.

IEC60079-20-1
LEL(vol%)

-

1.2

-

1.5

-

1.8

-

2.1

-

0.8

0.9

-

0.8

0.9

-

1.4

1.8

-

0.7

-

-

-

0.8

-

0.8

1.0

-

1.0

1.1

-

1.1

1.0

-

1.1

-

1.4

1.8

3. 可燃性ガスの危険性（Flammable Gas hazards）
3-4. 可燃性ガスの物性(Flammable gase date)NO.

CAS-No.

ｶﾞｽ名

99100-69-6

2-ﾋﾞﾆﾙﾋﾟﾘｼﾞﾝ

100103-09-3酢酸-2-ｴﾁﾙﾍｷｼ
ﾙ101

103-11-7ｱｸﾘﾙ酸-2-ｴﾁﾙﾍ
ｷｼﾙ

102

104-76-72-ｴﾁﾙﾍｷｻﾉｰﾙ

103

105-45-3

ｱｾﾄ酢酸ﾒﾁﾙ104105-46-4

sec-酢酸ﾌﾞﾁﾙ

105105-48-6ｸﾛﾛ酢酸ｲｿﾌﾟﾛﾋﾟ
ﾙ

106

105-54-4

酪酸ｴﾁﾙ

107

105-58-8ｼﾞｴﾁﾙｶｰﾎﾞﾈｰﾄ

108

106-35-4

ﾉﾙﾏﾙﾌﾞﾁﾙｴﾁﾙｹ
ﾄﾝ

109

106-42-3

ｷｼﾚﾝ

110

106-46-7

ｼﾞｸﾛﾛﾍﾞﾝｾﾞﾝ

111

106-58-1

1,4-ｼﾞﾒﾁﾙﾋﾟﾍﾟﾗ
ｼﾞﾝ

112

106-89-8

ｴﾋﾟｸﾛﾛﾋﾄﾞﾘﾝ

113

106-92-3

ｱﾘﾙｸﾞﾘｼｼﾙﾞｴｰﾃ
ﾙ114

106-96-7

ﾌﾟﾛﾊﾟｷﾞﾙﾌﾞﾛﾏｲﾄﾞ

115

106-97-8

ﾉﾙﾏﾙﾌﾞﾀﾝ

Name formula2-Vinylpyridine(= 2-Ethenylpyridine)(= α-Vinylpyridine)
NC(CH2=CH)CHCHCHCH
Acetic acid-2-ethylhexyl ester(= 2-Ethylhexyl acetate)
CH3COOCH2CH(C2H5)C4H9
Prop-2-enoic acid 2-ethylhexyl ester(= 2-Ethylhexyl 2-propenoate)(= 2-Ethylhexyl acrylate)
CH2=CHCOO(CH2)4CH3
2-Ethyl-1-hexanol
CH3 (CH2 )3CH(CH2CH3 )CH2OH
3-Oxo-butanoic acid methyl ester(= Acetoacetic acid methyl ester)(= 1-Methoxybutane-1,3-dione)(= Methyl acetoacetate)
CH3COOCH2COCH3
Acetic acid 1-methylpropyl ester(= sec-Butyl acetate)(= sec-Butyl ester of acetic acid)(= 1-Methylpropyl acetate)
CH3COOCH(CH3)CH2CH3
Chloroacetic acid-1-methylethyl ester(= iso-Propyl chloroacetate)(= Propan-2-yl 2-chloroacetate)
ClCH2COOCH(CH3)2
Butanoic acid ethyl ester(= Ethyl butanoate)(= Ethyl butyrate)(= Butyric acid ethyl ester)
CH3CH2CH2COO CH2CH3
Carbonic acid diethyl ester(= Diethyl carbonate)(CH3CH2O)2CO
3-Heptanone(= Ethyl butyl ketone)
CH3CH2CO[CH2]3CH3
1,4-Dimethyl benzene(= p-Xylene)(= p-Xyol)
C6H4(CH3)2
1,4-Dichlorobenzene(= Dichlorocide)
C6H4Cl2
1,4-Dimethylpiperazine
NH(CH3)CH2CH2NH(CH3)CH2CH2(Chloromethyl) oxirane(= Epichlorohydrin)(= 1-Chloro-2,3-epoxypropane)(= 2-Chloropropylene oxide)
OCH2 CHCH2Cl[(2-Propenyloxy) methyl] oxirane(= Allyl 2,3- epoxypropylether)(= 1-(Allyloxy)-2,3-epoxypropan)(= Glycidyl allyl ether)(= Allyl glycidyl ether)
CH2=CH-CH2-O-CHCH2CH2O
3-Bromo-1-propine(= Bromo propyne)
CH3CH≡CBrn-Butane(= Butyl hydride)(= Diethyl)(= Methylethylmethane)
CH3(CH2)2CH2- 15 -

理研計器
標準

ISO
10156 4th ed.

IEC60079-20-1
LEL(vol%)

-

1.2

-

0.8

-

0.7

-

0.9

-

1.3

-

1.3

-

1.6

-

1.4

-

1.4

-

1.1

-

0.9

1.0

-

2.2

2.2

-

1.0

-

2.3

-

-

-

3.0

1.4

1.4

0.6

1.4

2.3

1.5

3. 可燃性ガスの危険性（Flammable Gas hazards）
3-4. 可燃性ガスの物性(Flammable gase date)NO.

CAS-No.

ｶﾞｽ名

116106-98-9

1-ﾌﾞﾃﾝ117

106-99-0

ﾌﾞﾀｼﾞｴﾝ118

107-00-6

1-ﾌﾞﾁﾝ

119

107-02-8

ｱｸﾛﾚｲﾝ120

107-05-1

ｱﾘﾙｸﾛﾗｲﾄﾞ

121

107-06-2

1,2-ｼﾞｸﾛﾛｴﾀﾝ

122

107-07-3ｴﾁﾚﾝｸﾛﾛﾋﾄﾞﾘﾝ

123107-10-8

ﾉﾙﾏﾙﾌﾟﾛﾋﾟﾙｱﾐﾝ

124

107-13-1

ｱｸﾘﾛﾆﾄﾘﾙ

125

107-15-3ｴﾁﾚﾝｼﾞｱﾐﾝ

126

107-18-6

ｱﾘﾙｱﾙｺｰﾙ127

107-19-7

ﾌﾟﾛﾊﾟｷﾞﾙｱﾙｺｰﾙ128

107-20-0

ｸﾛﾛｱｾﾄｱﾙﾃﾞﾋﾄﾞ

Name formula
1-Butene(= n-Butylene)(= Ethylethylene)
CH2=CHCH2CH3
1,3-Butadiene(= Biethylene)(= Bivinyl)(= Divinyl)(= Erythrene)(= Vinylethylene)
CH2=CHCH=CH2
1-Butine(= Ethylacetylene)
CH3CH2C≡CH
2-Propenal (inhibited)(= Acraldehyde)(= Acrylaldehyde)(= Acrylic aldehyde)(= Allyl aldehyde)(= Propenal)(= Acrolein)
CH2=CHCHO
3-Chloro-1-propene(= Allyl chloride)(= 1-Chloro-2-propene)(= 3-Chloropropylene)
CH2=CHCH2Cl
1,2-Dichloroethane(= Ethylene chloride)(= Ethylene dichloride)
CH2ClCH2Cl
Ethylene chlorohydrin(= 2-Chloroethanol)(= 2-Chloroethyl alcohol)
CH2ClCH2OH
1-Propaneamine(= 1-Aminopropane)
CH3(CH2)2NH2
2-Propenenitrile(= Acrylonitrile)(= Cyanoethylene)(= Propenenitrile)(= Acrylonitrile)(= Vinyl cyanide, VCN)
CH2=CHCN
1,2-Ethanediamine(= Ethylenediamine)(= Dimethylenediamine)
NH2CH2CH2NH2
2-Propen-1-ol(= Allylic alcohol)(= Propenol)(= Allyl alcohol)(= Vinyl carbinol)
CH2=CHCH2OH
2-Propine-1-ol(= Prop-2-yn-1-ol)(= Propargyl alcohol)
HC≡CCH2OH
Chloroacetaldehyde(= 2-Chloroethanal)
ClCH2CHO- 16 -

ISO
10156 4th ed.

IEC60079-20-1
LEL(vol%)

理研計器
標準

1.5

1.6

1.6

1.4

1.4

1.1

1.3

-

-

2.8

2.8

-

2.9

2.9

-

6.2

6.2

-

4.9

4.9

-

2.0

2.0

-

2.8

2.8

-

2.5

2.7

-

2.5

2.5

-

2.4

-

5.7

3. 可燃性ガスの危険性（Flammable Gas hazards）
3-4. 可燃性ガスの物性(Flammable gase date)NO.

CAS-No.

ｶﾞｽ名

129107-30-2

ｸﾛﾛﾒﾁﾙﾒﾁﾙｴｰﾃ
ﾙ130

107-31-3ｷﾞ酸ﾒﾁﾙ131

108-01-0

N,N-ｼﾞﾒﾁﾙｴﾀﾉｰ
ﾙｱﾐﾝ

132

108-03-2

ﾆﾄﾛﾌﾟﾛﾊﾟﾝ

133

108-05-4

酢酸ﾋﾞﾆﾙ134

108-10-1

ﾒﾁﾙｲｿﾌﾞﾁﾙｹﾄﾝ

135

108-11-24-ﾒﾁﾙ-2-ﾍﾟﾝﾀﾉｰ
ﾙ

136

108-18-9

ｼﾞｲｿﾌﾟﾛﾋﾟﾙｱﾐﾝ

137

108-20-3

ｲｿﾌﾟﾛﾋﾟﾙｴｰﾃﾙ

138

108-21-4酢酸ｲｿﾌﾟﾛﾋﾟﾙ

139

108-24-7

無水酢酸

140

108-38-3

ｷｼﾚﾝ

141

108-62-3

ﾒﾀｱﾙﾃﾞﾋﾄﾞ

142

108-67-8ﾄﾘﾒﾁﾙﾍﾞﾝｾﾞﾝ

143

108-82-7

ｼﾞｲｿﾌﾞﾁﾙｶﾙﾋﾞﾉｰ
ﾙ

144

108-87-2ﾒﾁﾙｼｸﾛﾍｷｻﾝ

Name formula
Chloromethoxymethane(= Chloromethyl methyl ether)(= Chlorodimethyl ether)(= Chloromethoxy methane)(= Dimethylchloroether)(= Methylchloromethyl ether)
CH3OCH2Cl
Formic acid methyl ester(= Methyl formate)(= Methyl methanoate)
HCOOCH3
2-(Dimethylamino)ethanol(CH3)2NC2H4OH
1-Nitropropane
CH3CH2CH2NO2
Acetic acid ethenyl ester(= Vinyl acetate)(= 1-Acetoxyethylene)
CH3COOCH=CH2
4-Methylpentan-2-one(= Hexone)(= Isopropylacetone)(= Methyl isobutyl ketone)(CH3)2CHCH2COCH3
4-Methylpentan-2-ol(= Isobutylmethylcarbinol)(= Methyl amyl alcohol)(= Methyl isobutyl carbinol)(CH3)2CHCH2CHOHCH3n-(1-Methylethyl)-2-propanamine(= Diisopropylamine)((CH3)2CH)2NH
2,2´-Oxybispropane(= Diisopropyl ether)(= 2-Isopropoxy propane)((CH3)2CH)2O
Acetic acid-1-methylethyl ester(= iso-propyl acetate)(= iso-propyl ester of acetic acid)(= 1-Methylethyl ester of acetic acid)(= 2-Propyl acetate)
CH3COOCH(CH3)2
Acetic anhydride(= Acetic acid anhydride)(= Acetic oxide)(= Acetyl oxide)(= Ethanoic anhydride)(CH3CO)2O
1,3-Dimethylbenzene(= m-Xylene)(= m-Xylol)
C6H4(CH3)2
2,4,6,8-Tetramethyl-1,3,5,7tetraoxocane(= Metaldehyde)(C2H4O)4
1,3,5-Trimethylbenzene(= Mesitylene)
CHC(CH3)CHC(CH3)CHC(CH3)
2,6-Dimethylheptan-4-ol(= Diisobutylcarbinol)((CH3)2CHCH2)2CHOH
Methylcyclohexane(= Hexahydrodoluene)
CH3CH(CH2)4CH2- 17 -

理研計器
標準

ISO
10156 4th ed.

IEC60079-20-1
LEL(vol%)

-

-

5.0

5.0

4.5

-

-

1.5

-

2.2

-

2.6

2.6

-

1.2

1.2

-

1.14

1.0

-

1.2

-

1.0

1.4

-

1.7

1.8

-

2.0

2.0

-

1.0

1.0

-

-

-

0.8

-

0.7

-

1.0

1.1

1.15

3. 可燃性ガスの危険性（Flammable Gas hazards）
3-4. 可燃性ガスの物性(Flammable gase date)NO.

CAS-No.

ｶﾞｽ名

145108-88-3

ﾄﾙｴﾝ

146

108-89-44-ﾒﾁﾙﾋﾟﾘｼﾞﾝ

147

108-90-7

ﾓﾉｸﾛﾛﾍﾞﾝｾﾞﾝ

148

108-91-8

ｼｸﾛﾍｷｼﾙｱﾐﾝ

149108-93-0

ｼｸﾛﾍｷｻﾉｰﾙ

150

108-94-1

ｼｸﾛﾍｷｻﾉﾝ

151

108-95-2

ﾌｪﾉｰﾙ

152

108-99-63-ﾒﾁﾙﾋﾟﾘｼﾞﾝ

153

109-06-82-ﾒﾁﾙﾋﾟﾘｼﾞﾝ

154

109-55-7

3-ｼﾞﾒﾁﾙｱﾐﾉﾌﾟﾛﾋﾟ
ﾙｱﾐﾝ

155

109-60-4

酢酸ﾉﾙﾏﾙﾌﾟﾛﾋﾟﾙ

156

109-65-9

臭化ﾌﾞﾁﾙ157

109-66-0

ﾉﾙﾏﾙﾍﾟﾝﾀﾝ

158

109-69-3

塩化ﾌﾞﾁﾙ

159

109-73-9

ﾓﾉﾌﾞﾁﾙｱﾐﾝ

Name formula
Methyl benzene(= Toluene)(= Methyl benzol)(= Phenyl methane)
C6H5CH3
4-Methylpyridine(= γ-Picoline)
NCHCHC(CH3)CHCH2.
Chlorobenzene(= Phenyl chloride)(= Monochlorobenzene)
C6H5ClCyclohexylamine(= Aminocyclohexane)(= Aminohexahydro-benzene)(= Hexahydroaniline)(= Hexahydro-benzenamine)
CH2 (CH2)4CHNH2
Cyclohexanol(= Cyclohexyl alcohol)(= Hexahydrophenol)(= Hexalin)
CH2 (CH2)4CHOH
Cyclohexanone(= Anone)(= Cyclohexyl ketone)(= Pimelic ketone)
CH2 (CH2)4CO
Phenol(= Carbolic acid)(= Hydroxybenzene)(= Monohydroxybenzene)(= Monophenol)(= Oxybenzene)
C6H5OH
3-Methylpyridine(= β-Picoline)
NCHC(CH3)CHCHCH
2-Methylpyridine(= α-Picoline)
NC(CH3)CHCHCHCH.
N,N-Dimethylpropane-1,3-diamine(= 3-Dimethylamino-propylamine)(= 1-Amino-3-dimethyl-aminopropane)(CH3)2N(CH2)3NH2
Acetic acid n-propyl ester(= n-Propyl acetate)(= 1-Acetoxypropane)(= n-propyl ester acetic acid)
CH3COOCH2CH2CH3
1-Bromobutane(= n-Butyl bromide)
CH3(CH2)2 CH2Br
n-Pentane
CH3(CH2)3CH3
1-Chlorobutane(= n-Butyl chloride)(= n-Propylcarbinyl chloride)
CH3(CH2)2CH2Cl
1-Aminobutane(= n-Butylamine)
CH3(CH2)3NH2- 18 -

ISO
10156 4th ed.

IEC60079-20-1
LEL(vol%)

理研計器
標準

1.0

1.0

1.2

-

1.1

-

1.3

-

1.1

-

1.2

1.2

-

1.3

1.1

-

1.3

1.8

-

1.4

-

1.2

-

1.2

-

1.7

-

2.5

1.1

1.1

-

1.8

-

1.7

1.3

1.7

1.5

1.7

3. 可燃性ガスの危険性（Flammable Gas hazards）
3-4. 可燃性ガスの物性(Flammable gase date)NO.

CAS-No.

ｶﾞｽ名

160

109-79-5

ﾉﾙﾏﾙﾌﾞﾁﾙﾒﾙｶﾌﾟ
ﾀﾝ

161

109-86-4ｴﾁﾚﾝｸﾞﾘｺｰﾙﾓﾉﾒ
ﾁﾙｴｰﾃﾙ162

109-87-5

ｼﾞﾒﾄｷｼﾒﾀﾝ

163

109-89-7ｼﾞｴﾁﾙｱﾐﾝ

164

109-94-4

ギ酸ｴﾁﾙ

165

109-95-5 or(8013-58-9)
comment:
both
are
valid

亜硝酸ｴﾁﾙ166

109-99-9ﾃﾄﾗﾋﾄﾞﾛﾌﾗﾝ

167

110-00-9

ﾌﾗﾝ

168

110-01-0ﾃﾄﾗﾋﾄﾞﾛﾁｵﾌｪﾝ

169

110-02-1

ﾁｵﾌｪﾝ

170

110-05-4

ｼﾞ-ﾀｰｼｬﾙﾌﾞﾁﾙﾍﾟ
ﾙｵｷｼﾄﾞ

171

110-43-0ﾒﾁﾙﾉﾙﾏﾙｱﾐﾙｹﾄ
ﾝ

172

110-54-3(n-Hexane)

ﾉﾙﾏﾙﾍｷｻﾝ

Name formula
1-Butanethiol(= Butanethiol)(= n-Butyl mercaptan)(= n-Butanethiol)(= 1-Mercaptobutane)
CH3(CH2)3SH
2-Methoxyethanol(= Ethylene glycol monomethyl ether)
CH3OCH2CH2OH
Dimethoxymethane(= Methylal)(= Dimethyl acetal methanal)(= Dimethyl acetal formaldehyde)(= Dimethyl formal)(= 2,4-Dioxapentane)
CH2(OCH3)2n-Ethylethanamine(= Diethamine)(= Diethylamine)(C2H5)2NH
Formic acid ethyl ester(= Ethyl methanoate)(= Ethyl formate)
HCOOCH2CH3
Nitrous acid ethyl ester(= Ethyl nitrite ; see 5.2.2)
CH3CH2ONO
Tetrahydrofuran(= 1,4-Epoxybutane)(= Oxolane)(= Oxacyclopentane)(= Tetramethylene oxide)
CH2(CH2)2CH2O
Furan(= Divinylene oxide)(= Furfuran)(= Tetrole)(= Oxole)(= Oxacyclopentadiene)
CH=CHCH=CHO
Tetrahydrothiophene(= Tetramethylene sulphide)(= Thiolane)(= Thiophane)(= Thiocyclopentane)
CH2(CH2)2CH2S
Thiophene(= Divinylene sulphide)(=Thiacyclopentadiene)(= Thiofuran)
CH=CHCH=CHS
bis(1,1-Dimethylethyl) peroxide(= tert-Dibutyl peroxide)(CH3)3COOC(CH3)3
Heptan-2-one(= 1-Methylhexanal)(= 2-Oxoheptane)(= Amyl methyl ketone)(= Butylacetone)
CH3CO(CH2)4CH3
Hexane (mixed isomers)(= Hexyl hydride)
CH3(CH2)4CH3- 19 -

理研計器
標準

ISO
10156 4th ed.

IEC60079-20-1
LEL(vol%)

-

1.4

-

1.8

2.5

-

2.2

2.2

-

1.7

1.8

2.7

2.7

-

3.0

-

1.5

-

2.3

-

1.1

-

1.5

-

0.74

-

1.1

1.0

1.0

2.0

1.1

1.2

3. 可燃性ガスの危険性（Flammable Gas hazards）
3-4. 可燃性ガスの物性(Flammable gase date)NO.

CAS-No.

ｶﾞｽ名

173110-62-3

ﾊﾞﾚﾙｱﾙﾃﾞﾋﾄﾞ

174

110-71-4ｴﾁﾚﾝｸﾞﾘｺｰﾙｼﾞﾒ
ﾁﾙｴｰﾃﾙ175

110-80-5ｴﾁﾚﾝｸﾞﾘｺｰﾙﾓﾉｴ
ﾁﾙｴｰﾃﾙ176

110-82-7

ｼｸﾛﾍｷｻﾝ177

110-83-8

ｼｸﾛﾍｷｾﾝ

178

110-86-1

ﾋﾟﾘｼﾞﾝ179

110-88-3

1,3,5‐ﾄﾘｵｷｻﾝ

180110-91-8

ﾓﾙﾎﾘﾝ

181

110-96-3

ｼﾞｲｿﾌﾞﾁﾙｱﾐﾝ

182

111-15-9ｴﾁﾚﾝｸﾞﾘｺｰﾙﾓﾉｴ
ﾁﾙｴｰﾃﾙｱｾﾃｰﾄ

183

111-27-3

ﾉﾙﾏﾙﾍｷｼﾙｱﾙ
ｺｰﾙ

184

111-43-3

ｼﾞﾌﾟﾛﾋﾟﾙｴｰﾃﾙ

185

111-49-9

ﾍｷｻｴﾁﾚﾝｲﾐﾝ

186

111-65-9

ﾉﾙﾏﾙｵｸﾀﾝ

Name formula
1-Pentanal(= Amyl aldehyde)(= Butyl formal)(= Valeraldehyde)
CH3 (CH2 )3CHO
1,2-Dimethoxyethane(= Monoglyme)(= Ethylene glycol dimethyl ether)(= Dimethylglycol)(= 2,5-Dioxahexane)
CH3O(CH2)2OCH3
2-Ethoxyethanol(= Ethane-1,2-diol ethyl ether)(= Ethyl cellosolve)(=3-Oxapentan-1-ol)(= Ethylene glycol ethyl ether)(= Ethylene glycol monoethyl ether)
CH3CH2OCH2CH2OH
Cyclohexane(= Hexahydrobenzene)(= Hexamethylene)(= Hexanaphthene)
CH2(CH2)4CH2
Cyclohexene(= Benzene tetrahydride)(= Tetrahydrobenzene)
CH2(CH2)3CH=CH
Pyridine(= Azine)(= Azabenzene)
C5H5N
1,3,5-Trioxane(= Trioxymethylene)
OCH2OCH2OCH2Morpholine(= Diethylene imidoxide)(= Diethylene oximide)(= Tetrahydro-1,4-oxazine)
OCH2CH2NHCH2CH2
2-Methyl-n-(2-methylpropyl)-1propanamine(= Diisobutylamine)((CH3)2CHCH2)2NH
Acetic acid 2-ethoxy-ethyl ester(= 2-Ethoxyethyl acetate)(=
Ethyleneglycolmonoethyl
etheracetate)(= Glycol monoethyl ether acetate)
CH3COOCH2CH2OCH2CH3
1-Hexanol(= Amylcarbinol)(= Hexyl alcohol)(= 1-Hydroxyhexane)(= Pentylcarbinol)
CH3(CH2)4CH3
1,1´-Oxybispropane(= Dipropylether)(= 1-propoxy-propane)
CH3(CH2)2O
Hexahydro-1H-acepine(= Azepane)
CH2 (CH2)5NH
n-Octane
CH3(CH2)6CH3- 20 -

理研計器
標準

ISO
10156 4th ed.

IEC60079-20-1
LEL(vol%)

-

1.4

-

1.6

1.6

-

1.7

1.8

1.0

1.0

1.3

-

1.1

1.2

-

1.7

1.8

-

3.2

-

1.4

-

0.8

-

1.2

1.7

-

1.1

1.3

-

1.18

-

-

0.8

0.8

2.0

0.8

3. 可燃性ガスの危険性（Flammable Gas hazards）
3-4. 可燃性ガスの物性(Flammable gase date)NO.

CAS-No.

ｶﾞｽ名

187111-69-3

ｱｼﾞﾎﾟﾆﾄﾘﾙ

188

111-70-6

1-ﾍﾌﾟﾀﾉｰﾙ

189

111-76-2ｴﾁﾚﾝｸﾞﾘｺｰﾙﾓﾉ
ﾌﾞﾁﾙｴｰﾃﾙ

190

111-84-2

ﾉﾅﾝ

191

111-87-5

1-ｵｸﾀﾉｰﾙ

192

111-90-0

ｼﾞｴﾁﾚﾝｸﾞﾘｺｰﾙﾓ
ﾉｴﾁﾙｴｰﾃﾙ

193

112-07-2ｴﾁﾚﾝｸﾞﾘｺｰﾙﾓﾉ
ﾌﾞﾁﾙｴｰﾃﾙｱｾﾃｰ
ﾄ

194

112-30-1

1-ﾃﾞｶﾉｰﾙ

195

112-34-5

ｼﾞｴﾁﾚﾝｸﾞﾘｺｰﾙﾓ
ﾉﾌﾞﾁﾙｴｰﾃﾙ

196

112-41-4

1-ﾄﾞﾃﾞｶﾝ

197

112-58-3ｼﾞﾍﾞﾝｼﾞﾙｴｰﾃﾙ

198

115-07-1

ﾌﾟﾛﾋﾟﾚﾝ

199

115-10-6ｼﾞﾒﾁﾙｴｰﾃﾙ

200115-11-7

ｲｿﾌﾞﾁﾚﾝ

201

116-14-3

四ﾌｯ化ｴﾁﾚﾝ

202

121-44-8

ﾄﾘｴﾁﾙｱﾐﾝ

Name formula
Hexanedinitrile(= 1,4-Dicyanobutane)(= Adiponitrile)(= Tetramethylene cyanide)
NC(CH2 )4CN
Heptan-1-ol(= hexylcarbinol)(= heptyl alcohol)(= enanthic alcohol)(= 1-hydroxyheptane)
CH3(CH2)5CH2OH
2-Butoxyethanol(= Ethylene glycol monobutyl ether)(= Butyl cellosolve)(= Butylglykol)
CH3(CH2)3OCH2OH
Nonane(= Nonyl hydride)
CH3(CH2)7CH2
1-Octanol(= Caprylic alcohol)(= Heptyl carbinol)(= 1-Hydroxyoctane)(= n-Octyl alcohol)
CH3(CH2)6CH2OH
2- (2-Ethoxyethoxy) ethanol(= Diethylene glycol monoethyl ether)(= 3,6-Dioxaoctan-1-ol)
CH3CH2OCH2CH2OCH2CH2OH
2-Butoxyethanol acetate(=
Ethyleneglycolmonobutyl
etheracetate)
C4H9O(CH2)2OCOCH3
1-Decanol(= Decyl alcohol)
CH3(CH2)9OH
2-(2-Butoxyethoxy) ethanol(= Butyldiglykol)(= Diglycol monobutyl ether)
CH3(CH2)3OCH2CH2OCH2CH2OH
1-Dodecene
CH3(CH2)9CH=CH2
1,1´-Oxybishexane(= Dihexyl Ether)(CH3(CH2)5)2O
Propene(= Methylethylene)(= Propylene)
CH2=CHCH3
Oxybismethane(= Methyl ether)(= Dimethylether)(= Wood ether)(= Methoxymethane)(CH3)2O
2-Methylprop-1-ene(= 1,1-Dimethylethylene)(= Isobutylene)(= Isobutene)(= 2-Methylpropene)(CH3)2C=CH2
Tetrafluoroethylene
CF2=CF2
N,N-Diethylethanamine(= Triethylamine)(CH3CH2)3N- 21 -

理研計器
標準

ISO
10156 4th ed.

IEC60079-20-1
LEL(vol%)

-

1.70

-

0.9

-

1.1

1.1

0.7

0.7

0.7

-

0.9

-

1.3

1.2

-

0.9

0.8

-

0.7

-

0.85

-

0.6

-

-

1.8

2.0

2.0

2.7

2.7

3.0

1.6

1.6

1.8

10.5

10

10

-

1.2

1.2

3. 可燃性ガスの危険性（Flammable Gas hazards）
3-4. 可燃性ガスの物性(Flammable gase date)NO.

CAS-No.

ｶﾞｽ名

203121-69-7

N,N-ｼﾞﾒﾁﾙｱﾆﾘﾝ

204

123-05-72-ｴﾁﾙﾍｷｻﾅｰﾙ

205

123-38-6

ﾌﾟﾛﾋﾟｵﾝｱﾙﾃﾞﾋﾄﾞ

206

123-42-2ｼﾞｱｾﾄﾝｱﾙｺｰﾙ207

123-51-3ｲｿｱﾐﾙｱﾙｺｰﾙ208

123-54-6ｱｾﾁﾙｱｾﾄﾝ

209

123-63-7

ﾊﾟﾗｱﾙﾃﾞﾋﾄﾞ

210

123-72-8

ﾉﾙﾏﾙﾌﾞﾁﾙｱﾙﾃﾞﾋ
ﾄﾞ211

123-86-4

酢酸ﾌﾞﾁﾙ212

123-91-1

ｼﾞｵｷｻﾝ213

124-13-0

1-ｵｸﾀﾅｰﾙ

214

124-18-5(n-Decane)

ﾉﾙﾏﾙﾃﾞｶﾝ

215

124-40-3ｼﾞﾒﾁﾙｱﾐﾝ

216

126-99-8ｸﾛﾛﾌﾟﾚﾝ

217

138-86-3

d-ﾘﾓﾈﾝ218

140-88-5ｱｸﾘﾙ酸ｴﾁﾙ

219

141-32-2ｱｸﾘﾙ酸ﾉﾙﾏﾙﾌﾞﾁﾙName formula
N,N-Dimethylbenzeneamine(= N,N-Dimethylaniline)
C6H3(CH3)2NH2
2-Ethylhexanal(= 2-Ethylhexaldehyde)
CH3CH(CH2CH3)(CH2)3CHO
1-Propanal(= Propionic aldehyde)
CH3CH2CHO
4-Hydroxy-4-methylpenta-2-one(= Diacetone alcohol)(= 2-Methyl-2-pentanol-4-one)
CH3COCH2C(CH3)2OH
3-Methylbutan-1-ol(= Isoamyl alcohol)(CH3)2CH (CH2)2OH
Pentane-2,4-dione(= Acetylacetone)
CH3COCH2COCH3
2,4,6-Trimethyl-1,3,5-trioxane(=p-Acetaldehyde)(= Paracetaldehyde)(= Paraldehyde)
OCH(CH3)OCH(CH3)OCH(CH3)
1-Butanal(= Butyraldehyde)(= Butyl aldehyde)
CH3CH2CH2CHO
Acetic acid n-butyl ester(= n-Butyl acetate)(= n-Butyl ester of acetic acid)(= Butyl ethanoate)
CH3COOCH2(CH2)2CH3
1,4-Dioxane(= Diethylene dioxide)(= Diethylene ether)
OCH2CH2OCH2CH2
Octanal(= Octaldehyde)
CH3(CH2)6CHO
Decane(mixed isomers)
C10H22n-Methylmethanamine(= Dimethylamine)(CH3)2NH
2-Chloro-1,3-butadiene(= Chloroprene)
CH2=CClCH=CH2
1-Methyl-4-(1-methylethenyl)
cyclohexene
CH3CCHCH2CH(C(CH3)=CH2)CH2C
H2
2-Propenoic acid ethyl ester(= Acrylic acid ethyl ester)(= Ethyl acrylate)(= Ethyl propenoate)
CH2=CHCOOCH2CH3
2-Propenoic acid butyl ester (inhibited)(= n-Butyl acrylate)(= Butyl ester of acrylic acid)(= Butyl-2-propenoate)
CH2=CHCOOC4H9- 22 -

ISO
10156 4th ed.

IEC60079-20-1
LEL(vol%)

理研計器
標準

-

1.2

1.2

-

0.9

-

2.0

2.3

-

1.8

1.8

-

1.3

-

1.7

-

1.3

-

1.7

2.5

-

1.2

1.4

-

1.4

1.9

-

-

0.7

0.7

0.7

2.8

2.8

2.8

-

1.9

-

0.7

0.7

-

1.4

1.4

-

1.2

1.5

3. 可燃性ガスの危険性（Flammable Gas hazards）
3-4. 可燃性ガスの物性(Flammable gase date)NO.

CAS-No.

ｶﾞｽ名

220141-43-5

ﾓﾉｴﾀﾉｰﾙｱﾐﾝ

221

141-78-6

酢酸ｴﾁﾙ

222

141-79-7

ﾒｼﾁﾙｵｷｻｲﾄﾞ223

141-97-9

ｱｾﾄ酢酸ｴﾁﾙ

224

142-29-0

ｼｸﾛﾍﾟﾝﾃﾝ

225

142-82-5(n-Heptane)

ﾉﾙﾏﾙﾍﾌﾟﾀﾝ

226

142-84-7

ｼﾞﾉﾙﾏﾙﾌﾟﾛﾋﾟﾙｱﾐﾝ

227142-96-1

ｼﾞﾌﾞﾁﾙｴｰﾃﾙ

228

151-56-4

ｴﾁﾚﾝｲﾐﾝ229287-23-0

ｼｸﾛﾌﾞﾀﾝ

230

287-92-3

ｼｸﾛﾍﾟﾝﾀﾝ

231

291-64-5

ｼｸﾛﾍﾌﾟﾀﾝ

232

300-62-9

ｱﾝﾌｪﾀﾐﾝ233

350-57-2(1,1,2,2－ﾃﾄﾗﾌﾙ
ｵﾛｴﾄｷｼ)ﾍﾞﾝｾﾞﾝ

234

359-11-5

ﾄﾘﾌﾙｵﾛｴﾁﾚﾝ

235

420-46-2

ﾌﾛﾝ 143a

236

461-53-0ﾌﾞﾀﾝ酸ﾌﾙｵﾘﾄﾞ

237

463-58-1

硫化ｶﾙﾎﾞﾆﾙ

238

493-02-7trans-ﾃﾞｶﾋﾄﾞﾛﾅﾌ
ﾀﾚﾝ

239

504-60-9

1，3-ﾍﾟﾝﾀｼﾞｴﾝ

240507-20-0

ﾀｰｼｬﾙ塩化ﾌﾞﾁﾙ

Name formula
2-Aminoethanol(= Ethanolamine)(= beta-Aminoethyl alcohol)(= Ethylolamine)(= 2-Hydroxyethylamine)(= Monoethanolamine)
NH2CH2CH2OH
Acetic acid ethyl ester(= Ethyl acetate)(= Ethyl ethanoate)
CH3COOCH2CH3
4-Methylpent-3-en-2-one(= Mesityl oxide)(CH3)2CCHCOCH33-Oxobutanoic acid ethyl ester(= Acetoacetic acid ethyl ester)(= 1-Ethoxybutane-1,3-dione)(= Ethyl acetoacetate)
CH3COCH2COOCH2CH3
Cyclopentene
CH=CHCH2CH2CH.
Heptane (mixed isomers)
C7H16
n-Propyl-1-propanamine(= Dipropylamine)(CH3CH2CH2)2NH
1,1´-Oxybisbutane(= Dibutyl ether)(= 1-Butoxybutane)(CH3(CH2)3)2O
Ethylenimine(= Aminoethylene)(= Aziridine)
CH3CH2N
Cyclobutane(= Tertamethylene)
CH2(CH2)2CH2
Cyclopentane(= Pentamethylene)
CH2(CH2)3CH2
Cycloheptane
CH2(CH2)3CH2(+-)-α-Methylbenzeneethanamine(= Amphetamine)(=1-Phenylpropan-2-amine)
C6H5CH2CH(NH2)CH3
1,1,2,2-Tetrafluoroethoxybenzene
C6H5OCF2CF2H
Trifluoroethylene
CF2=CFH
1,1,1-Trifluoroethane(= Methylfluoroform)
CF3CH3
Butanoyl fluoride(= Butyryl fluoride)
CH3(CH2)2COF
Carbonyl sulfide
COStrans-Decahydronaphthalene
CH2(CH2)3CHCH(CH2)3CH2
Penta-1,3-diene(= Piperylene)
CH2=CH-CH=CH-CH3
2-Chloro-2-methylpropane(CH3)3CCl- 23 -

ISO
10156 4th ed.

IEC60079-20-1
LEL(vol%)

理研計器
標準

-

-

5.5

2.0

2.0

2.1

-

1.6

0.4

-

1.0

-

1.48

0.8

0.85

1.1

-

1.2

1.8

-

0.9

1.5

-

3.3

1.8

1.8

-

1.4

-

1.1

-

-

-

1.6

10.5

15.3

7.0

6.8

-

2.6

6.5

6.5

-

0.7

-

1.2

-

-

1.4

9.8

12.0

3. 可燃性ガスの危険性（Flammable Gas hazards）
3-4. 可燃性ガスの物性(Flammable gase date)NO.

CAS-No.

ｶﾞｽ名

241513-35-9ｲｿﾍﾟﾝﾃﾝ242

513-36-0

塩化ｲｿﾌﾞﾁﾙ

243

526-73-8

1,2,3-ﾄﾘﾒﾁﾙﾍﾞﾝ
ｾﾞﾝ

244

534-22-52-ﾒﾁﾙﾌﾗﾝ

245

536-74-3ﾌｪﾆﾙｱｾﾁﾚﾝ

246540-54-5

1-ｸﾛﾛﾌﾟﾛﾊﾟﾝ

247

540-59-0

1，2-ｼﾞｸﾛﾛｴﾁﾚﾝ

248540-67-0

ｴﾁﾙﾒﾁﾙｴｰﾃﾙ

249

540-84-1

ｲｿｵｸﾀﾝ250

540-88-5

酢酸ﾀｰｼｬﾙﾌﾞﾁﾙ

251542-92-7ｼｸﾛﾍﾟﾝﾀｼﾞｴﾝ

252

544-01-4ｲｿｱﾐﾙｴｰﾃﾙ

253554-14-32-ﾒﾁﾙﾁｵﾌｪﾝ

254

557-99-3

ﾌｯ化ｱｾﾁﾙ

255

563-47-3

3-ｸﾛﾛ-2-ﾒﾁﾙ-1ﾌﾟﾛﾍﾟﾝ

256

583-48-2

3，4-ｼﾞﾒﾁﾙﾍｷｻﾝ

257590-01-2

ﾌﾟﾛﾋﾟｵﾝ酸ﾉﾙﾏﾙ
ﾌﾞﾁﾙ

258

590-18-1

ｼｽ-２-ﾌﾞﾃﾝ

259

590-86-33-ﾒﾁﾙﾌﾞﾀﾅｰﾙ

260

591-78-6

2-ﾍｷｻﾉﾝ

Name formula
2-Methylbut-2-ene(= Amylene)(= Trimethylethylene)(CH3)2C=CHCH3
1-Chloro-2-methylpropane(CH3)2CHCH2Cl
1,2,3-Trimethylbenzene(= Hemimellitene)
CHCHCH(CH3) C(CH3)C(CH3)
2-Methylfuran
OC(CH3) CHCHCH
Phenylacetylene(= Ethynylbenzene)(= Phenyl ethyne)
C6H5C≡CH
1-Chloropropane
CH3CH2CH2Cl
1,2-Dichloroethene(= Acetylene dichloride)(= trans-Acetylene dichloride)(= sym-Dichloroethylene)
ClCH=CHCl
Ethyl methyl ether(= Methoxythane)
CH3OCH2CH3
2,2,4-Trimethylpentane(= iso-Butyltrimethyl methane)(= iso-Octane)(CH3)2CHCH2C(CH3)3
Acetic acid 1,1-dimethylethyl ester(= tert-Butyl acetate)(= tert-Butyl ester of acetic acid)
CH3COOC(CH3)3
1,3-Cyclopentadiene
CH2CH=CHCH=CH
1,1´-Oxybis(3-methylbutane)(= Diisopentylether)(= Di(3-methyl-1-butyl) ether)(=
3-Methyl-1-(3-methyl-butoxy)butane)(CH3)2CH(CH2)2O(CH2)2CH (CH3)2
2-Methylthiophene
SC(CH3) CHCHCH
Acetyl fluoride
CH3COF
3-Chloro-2-methyl-1-propene
CH2=C(CH3)CH2Cl
3,4-Dimethylhexane
CH3CH2CH(CH3)CH(CH3)CH2CH3
Propionic acid butyl ester(= Propanoic acid, butyl ester)(= Butyl propanoate)(= Butyl propionate)
C2H5COOC4H9
2-Butene (cis)CH3CH=CHCH3
3-Methylbutanal(= iso-Pentanal)(= iso-Valeraldehyde)(= 3-Methylbutyraldehyde)(CH3)2CHCH2CHO
2-Hexanone(= Hexan-2-one)(= Methyl butyl ketone)
CH3CO(CH2)3CH3- 24 -

ISO
10156 4th ed.

IEC60079-20-1
LEL(vol%)

理研計器
標準

-

1.3

1.5

-

2.0

-

0.8

-

1.4

-

-

-

2.4

-

9.7

2.0

2.0

1.0

0.7

-

1.3

-

-

-

1.27

-

1.3

-

5.6

-

2.1

-

0.8

-

1.0

1.5

1.6

-

1.3

-

1.2

0.8

3. 可燃性ガスの危険性（Flammable Gas hazards）
3-4. 可燃性ガスの物性(Flammable gase date)NO.

CAS-No.

ｶﾞｽ名

261591-87-7

酢酸ｱﾘﾙ

262

592-77-8

2-ﾍﾌﾟﾀﾝ

263

598-61-8ﾒﾁﾙｼｸﾛﾌﾞﾀﾝ

264623-36-9

ﾍｷｾﾅｰﾙ

265

624-83-9ｲｿｼｱﾝ酸ﾒﾁﾙ266625-55-8ｷﾞ酸ｲｿﾌﾟﾛﾋﾟﾙ

267

626-38-0

sec-酢酸ｱﾐﾙ

268628-63-7

ﾉﾙﾏﾙ酢酸ｱﾐﾙ

269629-14-1

ｴﾁﾚﾝｸﾞﾘｺｰﾙｼﾞｴ
ﾁﾙｴｰﾃﾙ

270

630-08-0

一酸化炭素

271

645-62-52-ｴﾁﾙﾍｷｾﾅｰﾙ

272

646-06-0

1，3-ｼﾞｵｷｿﾗﾝ

273

674-82-8

ｼﾞｹﾃﾝ

274

677-21-4

3,3,3-ﾄﾘﾌﾙｵﾛﾌﾟﾛ
ﾍﾟﾝ

275

693-65-2

ﾉﾙﾏﾙｱﾐﾙｴｰﾃﾙ

276

760-23-6

277

764-48-7

278

765-43-5

3，4-ｼﾞｸﾛﾛ-1-ﾌﾞﾃﾝ
２-ﾋﾄﾞﾛｷｼｴﾁﾙﾋﾞﾆ
ﾙｴｰﾃﾙ
ｼｸﾛﾌﾟﾛﾋﾟﾙﾒﾁﾙｹﾄ
ﾝ

ISO10156 4th ed.

IEC60079-20-1
LEL(vol%)

-

1.7

-

-

-

-

-

1.46

-

5.3

-

-

-

11.0

-

1.0

-

-

10.9

10.9

-

-

-

2.3

-

-

-

4.7

-

-

-

1.3

2-(Vinyloxy)ethanol--

1-Cyclopropyl ethanone(= acetylcyclopropane)(= Cyclopropyl methyl ketone)
CH2CH2CHCOCH3-

1.7

Name formula
Acetic acid-2-propenyl ester(= Acetoxypropene)(= Acetic acid, allyl ester)(= Allyl acetate)
CH2=CHCH2OOCCH3
Hept-2-ene
CH3(CH2)3CH=CHCH3
Methylcyclobutane
CH3CH(CH2)2CH2
2-Methylpent-2-enal
CH3CH2CHC(CH3)COH
Methylisocyanate(= Methyl ester of isocyanic acid)
CH3NCO
Formic acid-1-methylethyl ester(= iso-Propyl formate)(= Formic acid isopropyl ester)(= 1-Methylethyl formate)
HCOOCH(CH3)2
Acetic acid 1-methylbutyl ester(= sec-Amyl acetate)(= 1-Methylbutyl acetate)(= 2-Pentanol acetate)(= 2-Pentyl ester of acetic acid)
CH3COOCH(CH3)(CH2)2CH3
Acetic acid penthyl ester(= n-Amyl acetate)(= Amyl acetic ester)(= 1-Pentanol acetate)(= Pentyl Acetate)(= Pentyl ester of acetic acid)(= Primary amyl acetate)
CH3COO(CH2)4CH3
1,2-Diethoxyethane(= 3,6-Dioxaoctane)
CH3CH2O(CH2)2OCH2CH3
Carbon monoxide (water saturated air
at 18° C; see 5.2.3)
CO
2-Ethyl-2-hexenal(= Ethylpropylacrolein)
CH3CH(CH2CH3)=CH(CH2)2CH3
1,3-Dioxolane(= glycolformal)(= formaldehyde ethylene acetal)(= ethylene glycol formal)
OCH2CH2OCH2
4-Methylene-2-oxetanone(= Acetyl ketene)(= But-3-en-3-olide)(= Diketene)
CH2=CCH2C(O)O
3,3,3-Trifluoroprop-1-ene
CF3CH=CH2
1,1´-Oxybispentane(= Dipentylether)(CH3(CH2)4)2O
3,4-Dichlorobut-1-ene
CH2=CHCHClCH2Cl- 25 -

理研計器
標準

12.5

4.0

3. 可燃性ガスの危険性（Flammable Gas hazards）
3-4. 可燃性ガスの物性(Flammable gase date)NO.

CAS-No.

ｶﾞｽ名279814-68-6ｱｸﾘﾙ酸ｸﾛﾗｲﾄﾞ

280

872-05-9

ﾉﾙﾏﾙﾃﾞｾﾝ

281

920-46-7

ﾒﾀｸﾘﾙ酸ｸﾛﾗｲﾄﾞ

282

926-57-8

1，3-ｼﾞｸﾛﾛ-2-ﾌﾞﾃ
ﾝ

283

994-05-8

ﾀｰｼｬﾙｱﾐﾙﾒﾁﾙ
ｴｰﾃﾙ

284

1120-56-5ﾒﾁﾚﾝｼｸﾛﾌﾞﾀﾝ

285

1122-03-8ﾄﾘﾒﾁﾙｽﾃｱﾘﾙｱﾝ
ﾓﾆｳﾑｸﾛﾘﾄﾞ

286

1300-73-8

ｷｼﾘｼﾞﾝ

287

1319-77-3(o-Cresol)

ｸﾚｿﾞｰﾙ

288

1333-74-0

289

1498-64-2

水素
ｴﾁﾙｼﾞｸﾛﾛﾁｵﾎｽ
ﾌｧｰﾄ

290

1634-04-4ﾒﾁﾙﾀｰｼｬﾙﾌﾞﾁﾙ
ｴｰﾃﾙ

291

1640-89-7ｴﾁﾙｼｸﾛﾍﾟﾝﾀﾝ

292

1678-91-7ｴﾁﾙｼｸﾛﾍｷｻﾝ

293

1712-64-7

硝酸ｲｿﾌﾟﾛﾋﾟﾙ

294

1719-53-5ｼﾞｴﾁﾙｼﾞｸﾛﾛｼﾗﾝ

295

1738-25-6

296

2032-35-1

297

2426-08-6

298

2673-15-6

299

2993-85-3

300

3583-47-93-（ｼﾞﾒﾁﾙｱﾐﾉ）
ﾌﾟﾛﾋﾟｵﾆﾄﾘﾙ
ﾌﾞﾛﾓｱｾﾄｱﾙﾃﾞﾋﾄﾞ
ｼﾞｴﾁﾙｱｾﾀｰﾙ

ﾉﾙﾏﾙﾌﾞﾁﾙｸﾞﾘｼｼﾞ
ﾙｴｰﾃﾙ

2-ﾒﾁﾙ2,2,3,3,4,4,5,5ｵｸﾀﾌﾙｵﾛ-2-ﾍｷｻﾉｰﾙ
ｱｸﾘﾙ酸2,2,3,3,4,4,5,5,6,6,7,7-ﾄﾞﾃﾞｶﾌﾙｵ
ﾛﾍﾌﾟﾁﾙ
2，3-ﾋﾞｽ（ｸﾛﾛﾒﾁﾙ）ｵｷｼﾗﾝ

理研計器
標準

ISO
10156 4th ed.

IEC60079-20-1
LEL(vol%)

-

2.68

-

0.55

-

2.5

-

-

-

1.18

-

1.25

-

-

-

1.0

-

1.1

1.1

4.0

4.0

4.0

-

-

-

1.5

-

1.05

-

0.9

-

2.0

-

3.4

-

1.57

-

-

-

-

2,2,3,3,4,4,5,5-Octafluoro-1,1dimethylpentan-1-ol
H(CF2CF2)2C(CH3)2OH--

2,2,3,3,4,4,5,5,6,6,7,7Dodecafluoroheptyl methacrylate
CH2=C(CH3)COOCH2(CF2)6H-

1.6

1,4-Dichloro-2,3 Epoxybutane(= 2,3-bis(chloromethyl) oxirane)
CH2ClCH2CHCHOCH2Cl
.

-

1.9

Name formula
Acryloyl chloride(= Propenoyl chloride)(= Acrylic acid chloride)
CH2CHCOCl
1-Decene
CH2(CH2)8CH3
Methacryloyl chloride(= Methacrylic acid chloride)(= 2-Methyl-2-propenoyl chloride)
CH2CCH3COCl
1,3-Dichloro-2-butene
CH3CCI=CHCH2Cl
2-Methoxy-2-methyl-butane(= 1,1-Dimethylpropyl methyl ether)(= Methyl tert-pentyl ether)(CH3)2C(OCH3)CH2CH3
Methylenecyclobutane
C(=CH2)(CH2)2CH2
4,4,5-Trimethyl-1,3-dioxane
OCH2OCH(CH3)C(CH3)2CH2
Xylidenes (Mixture of isomers)(= Xylidine)
C6H3(CH3)2NH2
Cresol (mixed isomers)
CH3C6H4OH
Hydrogen H2
O-Ethyl phosphoro dichloridothioate
C2H5OPSCl2
2-Methoxy-2-methylpropane(= tert-Butyl methylether)(= Methyl tert-butylether)
CH3OC(CH3)3
Ethylcyclopentane
CH3CH2CH(CH2)3CH2
Ethylcyclohexane
CH3CH2CH(CH2)4CH2
Nitric acid-1-methylethyl ester(= iso-Propyl nitrate)(= Nitric acid isopropyl ester)(= Propane-2-nitrate)(CH3)2CHONO2
Dichlorodiethylsilane(= Diethyl-dichloro-silane)(C2H5)2SiCl2
3-(Dimethylamino) propiononitrile(CH3)2NHCH2CH2CN
2-Bromo-1,1-diethoxyethane(CH3CH2O)2CHCH2Br(Butoxymethyl)oxirane(= n-Butyl glycidil ether)(= Butyl 2,3- Epoxypropylether)(= 1,2-Epoxy-3-butoxypropane)(CH2)3OCH2
CH3CH2(CH2)3O CH2 CHCH2O- 26 -

1.6

0.9

3. 可燃性ガスの危険性（Flammable Gas hazards）
3-4. 可燃性ガスの物性(Flammable gase date)NO.

CAS-No.

IEC60079-20-1
LEL(vol%)

-

2.1

-

1.2

-

0.78

-

2.0

-

2.4

-

-

15.4

15

15.0

硫化水素

Hydrogen Sulfide(= Hydrosulfuric acid)(= Sewer gas)(= Sulfuretted hydrogen) H2S

-

4.0

4.0

-

1.4

-

0.8

0.8

-

0.7

0.7

-

1.2

-

-

-

0.9

-

-

-

1.3

-

-

-

7.0

-

1.1

-

-

301

4170-30-3

ｸﾛﾄﾝｱﾙﾃﾞﾋﾄﾞ

302

4806-61-5ｴﾁﾙｼｸﾛﾌﾞﾀﾝ

303

5870-82-6

1,1,3-ﾄﾘｴﾄｷｼﾌﾞﾀ
ﾝ

3045891-21-4

5‐ｸﾛﾛ-2-ﾍﾟﾝﾀﾉﾝ

305

7383-71-3

2,2,3,3-ﾃﾄﾗﾌﾙｵ
ﾛﾌﾟﾛﾋﾟﾙｱｸﾘﾗｰﾄ

306

7397-62-8

ﾋﾄﾞﾛｷｼ酢酸ﾌﾞﾁﾙ

3077664-41-7

ｱﾝﾓﾆｱ

308

7783-06-4

理研計器
標準

ISO
10156 4th ed.

ｶﾞｽ名Name formula
2-Butenal(= Crotonaldehyde)(= beta-Methyl acrolein)(= Propylene aldehyde)
CH3CH=CHCHO
Ethylcyclobutane
CH3CH2CH(CH2)2CH2
1,1,3-Triethoxybutane(CH3CH2O)2CHCH2CH(CH3CH2O)C
H3
5-Chloro-2-pentanone
CH3CO(CH2)3Cl
2,2,3,3-Tetrafluoropropyl acrylate(= Acrylic acid 2,2,3,3-tetrafluoropropyl ester)(= 2,2,3,3-Tetrafluoro propyl prop-2enoate)
CH2=CHCOOCH2CF2CF2H
Hydroxyacetic butylester(= Butyl glycolate)(= Butyl-2-hydroxyacetate)
HOCH2COO(CH2)3CH3
Ammonia (= Anhydrous ammonia)
NH3

309

8006-61-9

ｶﾞｿﾘﾝ

Gasoline(= Motor fuel)(= Natural gasoline)(= Petrol)

310

8006-64-2

ﾃﾚﾋﾟﾝ油Turpentine oil

ｹﾛｼﾝ

311

8008-20-6

312

17639-76-8

313

20260-76-8

314

25377-83-7

ｵｸﾃﾝ

315

25639-42-3ﾒﾁﾙｼｸﾛﾍｷｻﾉｰﾙ

316

26519-91-5

317

29553-26-2

31830525-89-4

ﾊﾟﾗﾎﾙﾑｱﾙﾃﾞﾋﾄﾞ

319

34590-94-8

ｼﾞﾌﾟﾛﾋﾟﾚﾝｸﾞﾘｺｰﾙ
ﾓﾉﾒﾁﾙｴｰﾃﾙ

32035158-25-9

ｶｶｵｱﾙﾃﾞﾋﾄﾞ

2－ﾒﾄｷｼﾌﾟﾛﾊﾟﾝ
酸ﾒﾁﾙ
2‐ﾒﾁﾙ-5-ﾋﾞﾆﾙﾋﾟ
ﾘｼﾞﾝ

ﾒﾁﾙｼｸﾛﾍﾟﾝﾀｼﾞｴ
ﾝ
2,2,3,3-ﾃﾄﾗﾌﾙｵ
ﾛ 1,1‐ｼﾞﾒﾁﾙﾌﾟﾛ
ﾊﾟﾉｰﾙ

Kerosene(= Diesel Oil No. 1)(= Fuel Oil No. 1)
Methyl-2-methoxypropionate
CH3CH(CH3O)COOCH3
2-Methyl-5-vinylpyridine
NC(CH3)CHCHC(CH2=CH)CH
Octene (mixed isomers)
C8H16
Methylcyclohexanol (mixed isomers)(= Hexahydromethyl phenol)(= Hexahydrocresol)
C7H13OH
Methylcyclopentadiene-1,3(CH3)C=CHCH=CHCH2
2,2,3,3-Tetrafluoro-1,1dimethylpropan- 1-ol
HCF2CF2C(CH3)2OH
Paraformaldehyde(= Polyoxymethylene)(= Polymerised formaldehyde)(= Formaldehyde polymer)
poly(CH2O)(2-Methoxymethylethoxy)propanol(= Dipropylene glycol monomethyl
ether)
H3COC3H6OC3H6OH
2-iso-Propyl-5-methylhex-2-enal-(=
2-Hexenal,
5-methyl-2-(1methylethyl))(CH3)2CH-C(CHO)CHCH2CH(CH3)2- 27 -

3. 可燃性ガスの危険性（Flammable Gas hazards）
3-4. 可燃性ガスの物性(Flammable gase date)
CAS-No.

ｶﾞｽ名

32145102-52-1

2,2,3,3-ﾃﾄﾗﾌﾙｵ
ﾛﾌﾟﾛﾋﾟﾙﾒﾀｸﾘﾗｰﾄ

322

68476-34-6

ﾃﾞｨｰｾﾞﾙ燃料

323

No CAS

1,ｸﾛﾛ－2,2,2‐ﾄﾘ
ﾌﾙｵﾛﾒﾁﾙ

324

No CAS

ｺｰｸ炉ｶﾞｽ

325

No CAS

ﾊﾞﾝｶｰ重油

Fuel oil-64-ﾒﾁﾙﾃﾄﾗﾋﾄﾞﾛﾋﾟﾗﾝ
2-ﾒ ﾁﾙ -3,5-ﾍｷ ｻ
ｼﾞｴﾝ-2-ｵｰﾙ

4-Methylenetetra-hydropyran
OCH2CH2C(=CH2) CH2CH2
2-Methylhexa-3,5-dien-2-ol
CH2=CHC=CHC(OH)(CH3)2
Water gas
Mixture of CO + H2

NO.

Name formula
2,2,3,3-Tetrafluoropropyl methacrylat(= 2,2,3,3-Tetrafluoro propyl
2methylprop-2-enoate)
CH2=C(CH2)COOCH2CF2CF2H
Diesel Oil No. 2(= Diesel fuel No. 2)(=Fuel Oil No. 2)
1-Chloro-2,2,2-trifluoroethyl
methylether
CF3CHClOCH3
Coke oven gas(see 5.2.1)

ISO10156 4th ed.

IEC60079-20-1
LEL(vol%)

-

1.9

-

0.6

-

8.0

-

-

-

-

-

1.5

-

-

-

-

理研計器
標準

326

No CAS

327

No CAS

328

No CAS

水性ｶﾞｽ

329
330
331
332
333
334
335
336
337
338
339
340
341
342
343
344
345
346
347

7784-42-1

ｱﾙｼﾝ

Arsine

3.9

-

5.1

74-83-9

臭化ﾒﾁﾙBromomethan

8.6

-

8.6

590-19-2

1,2‐ﾌﾞﾀｼﾞｴﾝ

1,2-Butadiene

1.4-

624-64-6

ﾄﾗﾝｽﾌﾞﾃﾝ

trans-Butene

1.5-75-68-3

ﾌﾛﾝ 142b

Chlorodifluoroethane (R142b)

6.3-

460-19-5

ｼｱﾉｹﾞﾝ

Cyanogen

3.9-7782-39

重水素

Deuterium

6.7

-

5.0

19287-45-7

ｼﾞﾎﾞﾗﾝ

Diborane

0.9

-

0.8

4109-96-0

ｼﾞｸﾛﾛｼﾗﾝ

Dichlorosilane

2.5

-

4.1

75-37-6

ﾌﾛﾝ 152a

Difluoroethane (R152a)

4.0

-

3.7

463-82-1ｼﾞﾒﾁﾙﾌﾟﾛﾊﾟﾝ

Dimethylpropane (neopentane)

1.3-

353-36-6

ﾌﾙｵﾛｴﾁﾚﾝ

Fluoroethane

3.8-7782-65-2

ｹﾞﾙﾏﾝGermane1.0-7783-07-5

ｾﾚﾝ化水素Hydrogen selenide

4.0-7783-06-4

硫化水素

Hydrogen sulfide

3.9-

624-91-9

亜硝酸ﾒﾁﾙ

Methyl nitrite

5.3-

992-94-9ﾓﾉﾒﾁﾙｼﾗﾝ

Methyl silane

1.3-

563-45-13‐ﾒﾁﾙ-1-ﾌﾞﾃﾝ

Methylbutene (3-methylbut-1-ene)

1.5-7803-51-2

ﾎｽﾌｨﾝ

Phosphine

1.6

-

348

463-49-0

ﾌﾟﾛﾊﾟｼﾞｴﾝｱﾚﾝ

Propadiene

1.9

-

349
350
351
352
353

7803-62-5

ｼﾗﾝ

Silane1.4-

993-07-7ﾄﾘﾒﾁﾙｼﾗﾝ

Trimethylsilane

1.3-

593-60-2

臭化ﾋﾞﾆﾙ

Vinyl bromide

5.6-

75-02-5

ﾌｯ化ﾋﾞﾆﾙ

Vinyl fluoride

2.9-

107-25-5ﾒﾁﾙﾋﾞﾆﾙｴｰﾃﾙ

Vinyl methyl ether

2.2

-

1.4

-

354

503-17-3

2-ﾌﾞﾁﾝ

Dimethylacetylene
crotonylene)

355
356
357
358
359
360

112-40-3

ﾉﾙﾏﾙﾄﾞﾃﾞｶﾝ

n-Dodecane

0.6-78-00-2ﾃﾄﾗｴﾁﾙ鉛

Lead tetraethyl (tetraethyllead)

1.8-

75-09-2ｼﾞｸﾛﾛﾒﾀﾝ

Methylene chloride (Dichloromethane)

13.0

-

13465-78-6

ｸﾛﾛｼﾗﾝ

Monochlorosilane

1.0-13463-39-3

ﾆｯｹﾙｶﾙﾎﾞﾆﾙ

Nickel carbonyl (tetracarbonylnickel)

0.9-

110-74-7

ギ酸ﾌﾟﾛピル

Propyl formate

2.1-- 28 -(2-butyne,

7.8

0.8
4.0

1.8

0.8

1.4

13.0

4. 毒性ガスの危険性(Toxic Gas hazards)
4-1. 毒性ガスの危険性

44.

毒性ガスの危険性(Toxic Gas
hazards)

4-1. 毒性ガスの危険性
様々な産業において、使用されたり、プロセスの副生ガスとして生成されたりするガスの中には極低濃度でも人体に対して重大な健康被害や命の危険につながる毒性をもったガスが存在します。
いくつかのガス種、硫化水素(H2S)やアンモニア(NH3)等は特異臭がするため、人間でもガスの存在に気付くことが出来ますが、健康上の被害を及ぼすような濃度の閾値(例えば H2S TLV-TWA：
１ppm ACGIH2018)に達しているかどうかは人の嗅覚では判断できません。
１ppm とは、例えば 1000L(＝1t、1m3)の巨大な水槽にたった一滴(1ｍL＝1g、1cc)の毒液を垂らしてよくかき混ぜた濃度と同じです。その一滴(1ppm)が仮に醤油であったとしたら、攪拌すれば見分けもつかないでしょうし、溶けた水を舐めたとしても当然味も感じることはできないことは容易に想像できるでしょう。気体と液体の違いがあるにせよ、多くの毒性ガスといわれる物質は色もついていなけ
れば、嗅いでも匂いがしません。
よく一般家庭でもストーブなどの不完全燃焼で中毒死の原因となる“一酸化炭素(CO)”も無色無臭の毒性ガスの一つですが、ガスの存在に気付かないまま中毒になったり、死に至ってしまったりする“Silent killer”と呼ばれることもあります。

- 29 -

4. 毒性ガスの危険性(Toxic Gas hazards)

4-2. 許容濃度(Toxic Gas)

4-2. 許容濃度(Toxic Gas)

ガス名称

英語名称（ACGIH）

CASNo.

化学式

アセトアルデヒド

Acetaldehyde

75-07-0

C2H4O

酢酸

Acetic acid64-19-7

無水酢酸

Acetic anhydride

アセトン
アセトニトリル

ACGIH 2023
TWA

STEL

C2H4O2

10ppm

15ppm

108-24-7

C4H6O31ppm

3ppm

Acetone67-64-1

C3H6O

250ppm500ppmAcetonitrile

75-05-8

C2H3N

20ppm

アセトフェノン

Acetophenone

98-86-2

C8H8O

10ppm

アクロレイン

Acrolein107-02-8

C3H4O

アクリル酸

Acrylic acid79-10-7

C3H4O22ppm

アクリロニトリル

Acrylonitrile

107-13-1

C3H3N

2ppm

アリルアルコール

Allyl alcohol

107-18-6

C3H6O

0.5ppmアリルクロライド

Aｌlyl chloride

107-05-1

C3H5CL

1ppm

アリルグリシジルエーテル

Allyl glycidyl ether(AGE)

106-92-3

C6H10O2

1ppm

アンモニア

Ammonia

7664-41-7

NH3

25ppm

アニリン

Aniline62-53-3

C6H7N

2ppm

スチビン

Antimony hydride7803-52-3

H3Sb

0.1ppmアルシン

Arsine7784-42-1

AsH3

0.005ppm

ベンゼン

Benzene71-43-2

C6H6

0.5ppm

塩化ベンジル

Benzyl chloride

100-44-7

C7H7CL

1ppm

三臭化ホウ素Boron tribromide

10294-33-4

BBr3

三フッ化ホウ素Boron trifluoride

7637-07-2

BF3

0.1ppm臭素

Bromine7726-95-6

Br20.1ppm

五フッ化臭素Bromine pentafluoride

7789-30-2

BrF50.1ppmノルマルプロピルブロマイド

1-Bromopropane

106-94-5

C3H7Br

0.1ppmブタジエン

1,3-Butadiene106-99-0

C4H6

2ppm

イソブタン

Butane, all isomers

75-28-5

C4H10

1000ppm

ノルマルブタン

Butane, all isomers

106-97-8

C4H10

1000ppm

ノルマルブチルアルコール

n-Butanol

71-36-3

C4H10O- 30 -

C
25ppm

0.1ppm2ppm
35ppm

2.5ppm
0.7ppm20ppm

0.7ppm
0.2ppm

4. 毒性ガスの危険性(Toxic Gas hazards)

4-2. 許容濃度(Toxic Gas)
ACGIH 2023

ガス名称

英語名称（ACGIH）

CASNo.

化学式

２－ブチルアルコール

sec-Butanol

78-92-2

C4H10O

100ppmターシャルブチルアルコール

tert-Butanol

75-65-0

C4H10O

100ppm１－ブテン

Butenes, all isomers
Isobutene

106-98-9

C4H8

250ppm

イソブチレン

Butenes, all isomers
Isobutene

115-11-7

C4H8

250ppm２－ブテン

Butenes, all isomers
Isobutene

107-01-7

C4H8

250ppm

エチレングリコールモノブチルエーテル

2-Butoxyethanol(EGBE)

111-76-2

C6H14O2

20ppm

2-Butoxyethyl acetate(EGBEA)

112-07-2

C8H16O3

20ppm

エチレングリコールモノブチルエーテルアセ
テート
酢酸ブチル

TWA

STEL

n-Butyl acetate

123-86-4

C6H12O2

50ppm

150ppm

酢酸ターシャルブチル

tert-Butyl acetate

540-88-5

C6H12O2

50ppm

150ppm

アクリル酸ノルマルブチル

n-Butyl acrylate141-32-2

C7H12O22ppm

モノブチルアミン

n-Butylamine

109-73-9

C4H11N

二酸化炭素

Carbon dioxide

124-38-9CO25000ppm

二硫化炭素

Carbon disulfide

75-15-0

CS2

1ppm

一酸化炭素

Carbon monoxide

630-08-0

CO

25ppm

四臭化炭素

Carbon tetrabromide

558-13-4

CBr40.1ppm

0.3ppm

四塩化炭素

Carbon tetrachloride

56-23-5

CCL4

5ppm

10ppm

フッ化カルボニル

Carbonyl fluoride

353-50-4

COF2

2ppm

5ppm

硫化カルボニル

Carbonyl sulfide

463-58-1

COS

5ppm

カテコール

Catechol

120-80-9

C6H6O25ppm

塩素Chlorine

7782-50-5CL20.1ppm

0.4ppm

二酸化塩素

Chlorine dioxide

10049-04-4

CLO2

0.1ppm

0.3ppm

三フッ化塩素Chlorine trifluoride

7790-91-2

CLF3

モノクロロベンゼン

Chlorobenzene

108-90-7

C6H5CL

10ppm

ブロムクロロメタン

Chlorobromomethane

74-97-5

CH2CLBr

200ppm

フロン２２

Chlorodifluoromethane

75-45-6

CHF2CL

1000ppm

クロロホルムChloroform

67-66-3

CHCL3

10ppm

フロン１１５Chloropentafluoroethane

76-15-3

C2F5CL

1000ppm

クロロピクリン

Chloropicrin

76-06-2

CNO2CL3

0.1ppm

クロロトルエンo-Chlorotoluene

95-49-8

C7H7CL50ppm- 31 -

C

5ppm
30,000ppm

0.1ppm

4. 毒性ガスの危険性(Toxic Gas hazards)

4-2. 許容濃度(Toxic Gas)
ACGIH 2023

ガス名称

英語名称（ACGIH）

CASNo.

化学式

クメン

Cumene98-82-8

C9H12

5ppm

シクロヘキサンCyclohexane

110-82-7

C6H12

100ppm

シクロヘキサノール

Cyclohexanol

108-93-0

C6H12O

50ppm

シクロヘキサノン

Cyclohexanone

108-94-1

C6H10O

20ppm

シクロヘキセン

Cyclohexene

110-83-8

C6H10

20ppm

シクロペンタンCyclopentane

287-92-3

C5H10

1000ppm

ジアセトンアルコール

Diacetone alcohol

123-42-2

C6H12O2

50ppm

ジボラン

Diborane

19287-45-7

B2H6

0.1ppmジクロロベンゼンo-Dichlorobenzene
p-Dichlorobenzene

95-50-1(ｵﾙﾄ)106-46-7(ﾊﾟﾗ)C6H4CL2

25ppm(ｵﾙﾄ)10ppm(ﾊﾟﾗ)

フロン１２Dichlorodifluoromethane

75-71-8

CF2CL2

1000ppm

１，１－ジクロロエタン

1,1-Dichloroethane

75-34-3

C2H4CL2

100ppm

シス１，２－ジクロロエチレン

1,2-Dichloroethylene, all isomers

156-59-2

C2H2CL2

200ppmトランス－１，２－ジクロロエチレン

1,2-Dichloroethylene, all isomers

156-60-5

C2H2CL2

200ppm

フロン２１

Dichlorofluoromethane

75-43-4

CHFCL210ppm

ジクロロメタン

Dichloromethane75-09-2

CH2CL250ppm

１，３－ジクロロプロペン

1,3-Dichloropropene

542-75-6

C3H4CL2

1ppm

フロン１１４

Dichlorotetrafluoroethane

76-14-2

C2F4CL2

1000ppm

ジシクロペンタジエン
シクロペンタジエンを含む

Dicyclopentadiene
Including Cyclopenadien

77-73-6

C10H12

0.5ppm1ppm

ジエチルアミン

Diethylamine

109-89-7

C4H11N

5ppm

15ppm

ジエチレングリコールモノブチルエーテル

Diethylene glycol monobutyl ether

112-34-5

C8H18O310ppm

ジエチルケトン

Diethyl ketone

96-22-0

C5H10O

200ppmジブロモジフルオロメタン

Difluorodibromomethane

75-61-6

CF2Br2

100ppmジイソブチルケトン

Diisobutyl ketone

108-83-8

C9H18O

25ppm

ジイソプロピルアミン

Diisopropylamine

108-18-9

C6H15N

5ppm

Ｎ，Ｎ－ジメチルアセトアミド

N,N-Dimethyl acetamide

127-19-5

C4H9NO

10ppm

ジメチルアミン

Dimethylamine

124-40-3

C2H7N

5ppm

15ppm

Ｎ，Ｎ－ジメチルアニリン

Dimethylaniline

121-69-7

C8H11N

5ppm

10ppm

二硫化ジメチル

Dimethyl disulfide

624-92-0

C2H6S2

0.5ppmＮ，Ｎ－ジメチルホルムアミド

Dimethylformamide

68-12-2

C3H7NO

5ppm

ジメチル硫酸

Dimethyl sulfate

77-78-1

C2H6O4S

0.1ppm- 32 -

TWA

STEL

50ppm

50ppm(ｵﾙﾄ)300ppmC

4. 毒性ガスの危険性(Toxic Gas hazards)

4-2. 許容濃度(Toxic Gas)
ACGIH 2023

ガス名称

英語名称（ACGIH）

CASNo.

化学式

硫化ジメチルDimethyl sulfide

75-18-3

C2H6S

10ppm

ジオキサン

1,4-Dioxane

123-91-1

C4H8O220ppm

１，３－ジオキソラン

1,3-Dioxolane

646-06-0

C3H6O220ppm

ジビニルベンゼン

Divinyl benzene

1321-74-0

C10H1010ppm

エンフルレン

Enflurane

13838-16-9

C3H2ClF5O

75ppm

エピクロロヒドリン

Epichlorohydrin

106-89-8

C3H5OCL

0.5ppmエチルアルコール

Ethanol

64-17-5

C2H6O

モノエタノールアミン

Ethanolamine

141-43-5

C2H7NO

3ppm

エチレングリコールモノエチルエーテル

2-Ethoxyethanol(EGEE)

110-80-5

C4H10O25ppm

エチレングリコールモノエチルエーテルアセ
テート

2-Ethoxyethyl acetate(EGEEA)

111-15-9

C6H12O3

5ppm

酢酸エチル

Ethyl acetate141-78-6

C4H8O2

400ppm

アクリル酸エチル

Ethyl acrylate

140-88-5

C5H8O2

5ppm

15ppm

モノエチルアミン

Ethylamine

75-04-7

C2H7N

5ppm

15ppm

エチルベンゼン

Ethyl benzene

100-41-4

C8H10

20ppm

エチルブロマイド

Ethyl bromide

74-96-4

C2H5Br5ppm

エチルターシャルブチルエーテル

Ethyｌ tert-butyl ether

637-92-3

C6H14O

25ppm

モノクロロエタン

Ethyl chloride

75-00-3

C2H5CL

100ppmエチレン

Ethylene

74-85-1

C2H4200ppm

エチレンクロロヒドリン

Ethylene chlorohydrin

107-07-3

C2H5OCL

エチレンジアミン

Ethylenediamine

107-15-3

C2H8N210ppm

１，２－ジクロロエタン

Ethylene dichloride

107-06-2

C2H4CL210ppm

酸化エチレン

Ethylene oxide75-21-8

C2H4O

1ppm

エチレンイミン

Ethyleneimine

151-56-4

C2H5N

0.05ppm

0.1ppm

ジエチルエーテル

Ethyl ether

60-29-7

C4H10O

400ppm500ppm

ギ酸エチル

Ethyl formate

109-94-4

C3H6O2エチルデンノルボネン

Ethylidene norbornene

16219-75-3

C9H12

2ppm

エチルメルカプタン

Ethyl mercaptan

75-08-1

C2H6S

0.5ppm

テトラエトキシシラン

Ethyl silicate

78-10-4

C8H20O4Si

10ppm

フッ素

Fluorine

7782-41-4

F20.1ppmホルマリン

Formaldehyde

50-00-0CH2O

0.1ppm

0.3ppmホルムアルデヒド

Formaldehyde

50-00-0CH2O

0.1ppm

0.3ppm

ギ酸

Formic acid64-18-6

CH2O2

5ppm

10ppm

- 33 -

TWA

STEL

C

1000ppm
6ppm

1ppm

100ppm4ppm

0.5ppm

4. 毒性ガスの危険性(Toxic Gas hazards)

4-2. 許容濃度(Toxic Gas)
ACGIH 2023

ガス名称

英語名称（ACGIH）

CASNo.

化学式

フルフラールFurfural

98-01-1

C5H4O2

0.2ppmフルフリルアルコール

Furfuryl alcohol

98-00-0

C5H6O2

0.2ppm

ガソリン

Gasoline

68606-10-0
86290-81-5

ゲルマンGermanium tetrahydride7782-65-2

GeH40.2ppmハローセン

Halothane

151-67-7

C2HF3CLBr

50ppm

ノルマルヘプタン

Heptane, all isomers

142-82-5

C7H16

400ppmヘキサフルオロアセトン

Hexafluoroacetone684-16-2
34202-69-2(3 水和物)

C3OF6

0.1ppm

六フッ化プロピレン

Hexafluoropropylene

116-15-4

C3F6

0.1ppmノルマルヘキサン

n-Hexane

110-54-3

C6H14

50ppm

２，２－ジメチルブタン

Hexane isomers, other than n-Hexane

75-83-2

C6H14

500ppm１－ヘキセン

1-Hexene

592-41-6

C6H12

50ppm

2ppm

ヒドラジン

Hydrazine

302-01-2

N2H4

0.01ppm

2ppm

臭化水素

Hydrogen bromide

10035-10-6

HBr

2ppm

塩化水素

Hydrogen chloride

7647-01-0

HCL

2ppm

シアン化水素Hydrogen cyanide

74-90-8

CHN

フッ化水素

Hydrogen fluoride, as F

7664-39-3

HF

0.5ppm

過酸化水素

Hydrogen peroxide7722-84-1

H2O2

1ppm

セレン化水素Hydrogen selenide, as Se

7783-07-5

H2Se

0.05ppm

硫化水素

Hydrogen sulfide

7783-06-4

H2S

1ppm

5ppm

ヨウ素

Iodine7553-56-2

I2

0.01ppm

0.1ppm鉄カルボニルIron pentacarbonyl

13463-40-6

C5O5Fe0.1ppm

0.2ppm

イソブチルアルコール

Isobutanol

78-83-1

C4H10O

50ppm

イソホロン

Isophorone

78-59-1

C9H14O

イソプロピルアミン

Isopropylamine

75-31-0

C3H9N

5ppm

10ppm

イソプロピルエーテル

Isopropyl ether

108-20-3

C6H14O

250ppm

310ppm

メシチルオキサイド

Mesityl oxide

141-79-7

C6H10O

15ppm

25ppm

メタクリル酸Methacrylic acid79-41-4

C4H6O220ppm

メチルアルコール

Methanol

67-56-1

CH4O

200ppm

エチレングリコールモノメチルエーテル

2-Methoxyethanol(EGME)

109-86-4

C3H8O2

0.1ppm- 34 -

TWA

300ppmSTEL

C

500ppm500ppm1000ppm

4.7ppm2ppm

5ppm

250ppm

4. 毒性ガスの危険性(Toxic Gas hazards)
ガス名称
エチレングリコールモノメチルエーテルアセ
テート
メチルプロピレングリコール

4-2. 許容濃度(Toxic Gas)
ACGIH 2023

英語名称（ACGIH）

CASNo.

化学式

2-Methoxyethyl acetate(EGMEA)

110-49-6

C5H10O3

0.1ppmTWA

STEL

1-Methoxy-2-propanol

107-98-2

C4H10O2

50ppm

100ppmプロピレングリコールモノメチルエーテル

1-Methoxy-2-propanol

107-98-2

C4H10O2

50ppm

100ppmＰＭソルベント

1-Methoxy-2-propanol

107-98-2

C4H10O2

50ppm

100ppm

酢酸メチル

Methyl acetate79-20-9

C3H6O2

200ppm

250ppm

メチルアセチレン

Methyl acetylene

74-99-7

C3H4

1000ppm

アクリル酸メチル

Methyl acrylate

96-33-3

C4H6O22ppm

メタクリロニトリル

Methylacrylonitrile

126-98-7

C4H5N

1ppm

ジメトキシメタン

Methylal109-87-5

C3H8O2

1000ppm

モノメチルアミン

Methylamine

74-89-5

CH5N

5ppm

メチルアミルケトン

Methyl n-amyl ketone

110-43-0

C7H14O

50ppm

臭化メチル

Methyl bromide

74-83-9

CH3Br

1ppm

メチルターシャルブチルエーテル

Methyl tert-butyl ether(MTBE)

1634-04-4

C5H12O

50ppm

メチルノルマルブチルケトン

Methyl n-butyl ketone591-78-6

C6H12O

5ppm

10ppm

塩化メチル

Methyl chloride

74-87-3

CH3CL

50ppm

100ppm

１，１，１－トリクロロエタン

Methyl chloroform

71-55-6

C2H3CL3

350ppm450ppm

メチルシクロヘキサン

Methyl cyclohexane

108-87-2

C7H14

400ppm

メチルエチルケトン

Methyl ethyl ketone(MEK)

78-93-3

C4H8O

200ppm300ppm

ギ酸メチル

Methyl formate

107-31-3

C2H4O2

50ppm

100ppmモノメチルヒドラジン

Methyl hydrazine

60-34-4

CH6N2

0.01ppm

ヨウ化メチル

Methyl iodide

74-88-4

CH3I

2ppm

メチルイソアミルケトン

Methyl isoamyl ketone

110-12-3

C7H14O

20ppm

50ppm

４－メチル－２－ペンタノール

Methyl isobutyl carbinol

108-11-2

C6H14O

25ppm

40ppm

メチルイソブチルケトン

Methyl isobutyl ketone108-10-1

C6H12O

20ppm

75ppm

メチルイソプロピルケトン

Methyl isopropyl ketone563-80-4

C5H10O

20ppm

メチルメルカプタン

Methyl mercaptan

74-93-1

CH4S

0.5ppmメタクリル酸メチル

Methyl methacrylate

80-62-6

C5H8O250ppm

テトラメトキシシラン

Methyl silicate

681-84-5

C4H12O4Si

1ppm

α－メチルスチレン

α-Methyl styrene

98-83-9

C9H10

10ppm

メチルビニルケトン

Methyl vinyl ketone

78-94-4

C4H6O

モルホリン

Morpholine

110-91-8

C4H9NO

20ppm

ナフタレン

Naphthalene

91-20-3

C10H8

10ppm

- 35 -

C

15ppm

100ppm0.01ppm

4. 毒性ガスの危険性(Toxic Gas hazards)

4-2. 許容濃度(Toxic Gas)

ガス名称

英語名称（ACGIH）

CASNo.

化学式

硝酸

Nitric acid7697-37-2

一酸化窒素

Nitric oxide

10102-43-9

ニトロベンゼン

Nitrobenzene

ニトロエタン

Nitroethane

二酸化窒素
三フッ化窒素ACGIH 2023
TWA

STEL

HNO3

2ppm

4ppm

NO

25ppm

98-95-3

C6H5NO2

1ppm

79-24-3

C2H5NO2

100ppmNitrogen dioxide

10102-44-0NO20.2ppmNitrogen trifluoride

7783-54-2

NF3

10ppm

２－ニトロプロパン

2-Nitropropane

79-46-9

C3H7NO2

10ppm

一酸化二窒素Nitrous oxide

10024-97-2

N2O

50ppm

ノナン

Nonane111-84-2

C9H20

200ppm

イソオクタン

Octane, all isomers

540-84-1

C8H18

300ppmノルマルオクタン

Octane, all isomers

111-65-9C8H18

300ppm

四酸化オスミウム

Osmium tetroxide, as Os

20816-12-0OsO40.0002ppm

オゾン

Ozone

10028-15-6

O3

0.05ppm(Heavy)
0.08ppm(Moderate)
0.1ppm(Light)
0.2ppm(≤2hours)

イソペンタン

Pentane, all isomers

78-78-4

C5H12

1000ppm

ノルマルペンタン

Pentane, all isomers

109-66-0

C5H12

1000ppm

２，４－ペンタンジオン

2,4-Pentanedione123-54-6

C5H8O225ppm

酢酸イソアミル

Pentyl acetate, all isomers

123-92-2

C7H14O2

50ppm

フェノール

Phenol108-95-2

C6H6O

5ppm

ベンゼンチオール

Phenyl mercaptan

108-98-5

C6H6S

0.1ppmホスゲン

Phosgene75-44-5

COCL2

ホスフィン

Phosphine

7803-51-2

PH3

0.05ppm

オキシ塩化リン

Phosphorus oxychloride

10025-87-3

POCL3

0.1ppm

五塩化リン

Phosphorus pentachloride

10026-13-8

PCL50.1ppm

三塩化リン

Phosphorus trichloride

7719-12-2

PCL3

0.2ppmノルマルプロピルアルコールn-Propanol(n-Propyl alcohol)

71-23-8

C3H8O

100ppm

イソプロピルアルコール

2-Propanol

67-63-0

C3H8O

200ppmプロピオンアルデヒド

Propionaldehyde

123-38-6

C3H6O

20ppm

プロピオン酸Propionic acid79-09-4

C3H6O2

10ppm- 36 -

C

0.0006ppm

100ppm

0.02ppm
0.15ppm

0.5ppm
400ppm

4. 毒性ガスの危険性(Toxic Gas hazards)

4-2. 許容濃度(Toxic Gas)

ガス名称

英語名称（ACGIH）

CASNo.

化学式

酢酸ノルマルプロピル

n-Propyl acetate109-60-4

プロピレン

Propylene

115-07-1

１，２－ジクロロプロパン

Propylene dichloride

酸化プロピレン

Propylene oxide

プロピレンイミン
ピリジン

ACGIH 2023
TWA

STEL

C5H10O2

100ppm150ppmC3H6

500ppm

78-87-5

C3H6CL2

10ppm

75-56-9

C3H6O

2ppm

Propyleneimine

75-55-8

C3H7N

0.2ppmPyridine

110-86-1

C5H5N

1ppm

シラン

Silicon tetrahydride7803-62-5

SiH4

5ppm

スチレン

Styrene, monomer

100-42-5

C8H8

10ppm

二酸化硫黄

Sulfur dioxide

7446-09-5

SO2

六フッ化硫黄Sulfur hexafluoride

2551-62-4SF6四フッ化硫黄Sulfur tetrafluoride

7783-60-0SF4フッ化スルフリル

Sulfuryl fluoride

2699-79-8

SO2F2

5ppm

テトラブロモエタン

1,1,2,2,-Tetrabromoethane

79-27-6

C2H2Br4

0.1ppm

フロン１１２

1,1,2,2-Tetrachloro-1,2-difluoroethane

76-12-0

C2F2CL4

50ppm

１，１，２，２－テトラクロロエタン

1,1,2,2-Tetrachloroethane

79-34-5

C2H2CL4

1ppm

パークロロエチレン

Tetrachloroethylene

127-18-4

C2CL4

25ppm

四フッ化エチレン

Tetrafluoroethylene

116-14-3

C2F4

2ppm

テトラヒドロフラン

Tetrahydrofuran

109-99-9

C4H8O

50ppm

塩化チオニール

Thionyl chloride

7719-09-7

SOCL2

トルエン

Toluene108-88-3

C7H8

トリクロロベンゼン

1,2,4-Trichlorobenzene

120-82-1

C6H3Cl3

トリクロロエチレン

Trichloroethylene

79-01-6

C2HCL3

フロン１１

Trichlorofluoromethane

75-69-4

CFCL3

トリクロロプロパン

1,2,3-Trichloropropane

96-18-4

C3H5CL3

0.005ppm

フロン１１３

1,1,2-Trichloro-1,2,2-trifluoroethane

76-13-1

C2F3CL3

1000ppm

1250ppm

トリエチルアミン

Triethylamine

121-44-8

C6H15N

0.5ppm1ppm

フロン１３Ｂ１

Trifluorobromomethane

75-63-8

CF3Br

1000ppm

トリメチルアミン

Trimethylamine

75-50-3

C3H9N

5ppm

C9H12

10ppm

C3H9O3P

2ppm

C10H1620pm

トリメチルベンゼン

Trimethyl benzene(mixed isomers)

トリメチルホスファイト

Trimethyl phosphite

108-67-8
25551-13-7(異性体混合物)
121-45-9

テレピン油

Turpentine and selected monoterpenes

8006-64-2- 37 -

C

0.4ppm20ppm
0.25ppm

1000ppm
0.1ppm10ppm

100ppm100ppm
0.2ppm

20ppm
5ppm
10ppm

25ppm
1000ppm

15ppm

4. 毒性ガスの危険性(Toxic Gas hazards)

4-2. 許容濃度(Toxic Gas)
ACGIH 2023

ガス名称

英語名称（ACGIH）

CASNo.

化学式

α－ピネン

Turpentine and selected monoterpenes

80-56-8

C10H1620ppm

ペンタナールn-Valeraldehyde

110-62-3

C5H10O

50ppm

酢酸ビニル

Vinyl acetate

108-05-4

C4H6O2

10ppm

塩化ビニル

Vinyl chloride

75-01-4

C2H3CL

1ppm

ビニルシクロヘキセン

4-Vinyl cyclohexene

100-40-3

C8H12

0.1ppmＮ－ビニル－２ピロリドン

N-Vinyl-2-pyrrolidone

88-12-0

C6H9NO

0.05ppm

１，１－ジクロロエチレン

Vinylidene chloride

75-35-4

C2H2CL2

5ppm

フッ化ビニリデンモノマー

Vinylidene fluoride

75-38-7

C2H2F2

500ppmキシレン

Xylene(o, m & p isomers)

1330-20-7

C8H10

20ppm

オルトキシレン

Xylene(o, m & p isomers)

95-47-6(ｵﾙﾄ)

C8H10

20ppm

メタキシレン

Xylene(o, m & p isomers)

108-38-3(ﾒﾀ)

C8H10

20ppm

パラキシレン

Xylene(o, m & p isomers)

106-42-3(ﾊﾟﾗ)

C8H10

20ppm

- 38 -

TWA

STEL

15ppm

C

5. 酸素欠乏の危険性（Oxygen deficiency）
5-1. 酸素欠乏

55.

酸素欠乏の危険性（Oxygen
deficiency）

5-1. 酸素欠乏
酸素は人間の生命活動の維持には欠かせない物質であり、酸素欠乏症は脳をはじめとした人体に多大な影響を及ぼし、労働現場においても致死率の高い非常に危険な疾患の一つです日本国
内におけるこれらの労働災害の発生状況を業種別みると、そのほとんどが製造業、建設業で発生しており、毎年数名～数 10 名の被災者が発生しております。
日本の労働安全衛生法 酸素欠乏症等防止規則では『空気中の酸素濃度が 18％未満である状態』を酸素欠乏状態といいます。ガス検知器は 18％以上酸素があることを確認するために使用します。
酸素欠乏症の症状- 39 -

5. 酸素欠乏の危険性（Oxygen deficiency）
5-2. 酸素欠乏の主な 3 つの要因
各国の酸素欠乏閾値地域

レギュレーション

酸欠閾値

日本

労働安全衛生法

18.0％

酸素欠乏症等防止規則
アメリカ

OSHA

19.5％

労働安全衛生規則
ヨーロッパ(イギリス)

CoGDEM

19.5%※ OSHA（Occupational Safety and Health Administration ）労 働 安 全 衛 生 庁※ CoGDEM：イギリスガス検知器協議会

5-2. 酸素欠乏の主な 3 つの要因
1. 空気中の酸素の消費
酸素を消費する要因： 鉄などの金属の酸化(砂鉄、金属配管、金属製タンク)、
塗料の酸化、生物的な酸素消費(人間、微生物の呼吸)
2. 酸素欠乏空気の噴出や流入
様々な要因で発生した酸素欠乏空気が、作業内容、工法、気象条件によって酸素欠乏場
所に噴出、流入すると酸素欠乏症の要因になる。
3. メタン等の発生、不活性ガスの流入
自然界に存在するメタンの噴出や、製造業で使用する窒素、二酸化炭素、アルゴン等の不活性ガスのタンクや配管からの漏出が酸素欠乏になる

5-3. 過剰酸素
酸素は人間の生命活動にとって必須のものですが、高濃度または高分圧の酸素を摂取し続けると、全身の痙攣や意識喪失といった酸素中毒という症状になり、最悪の場合は死にいたるケー
スもあります。酸素が過剰な状態になりうる環境では、酸素欠乏(18％未満)だけでなく、高濃度になり過ぎないようにガス濃度を監視する必要があります。

- 40 -

6. ガス検知器の用途・設置場所(Typical areas that require gas detection)
6-1. ガス検知器の市場

66.

ガス検知器の用途・設置場所(Typical areas that requiregas detection)
6-1. ガス検知器の市場
ガスを使用する市場は全てガス検知器の市場になります。

1.研究所 / 大学 / 病院
可燃性ガスから毒性ガスまで様々なガスを使用する研究施設では、研究者の安全を守るため、当社の定置式ガス検知器によりガス漏洩を監視し、万一の際に早期検知をめざすといった対策が採られています。
また、その場で X 線回折（XRD）、蛍光 X
線（XRF）の 2 つの分析が可能な分析装置など、ガス検知器以外の製品も、持ち出すことのできない文化財等の研究・調査の場面で活躍しています。

- 41 -

6. ガス検知器の用途・設置場所(Typical areas that require gas detection)
6-1. ガス検知器の市場

2.エレクトロニクス産業
半導体/液晶パネル工場ではシラン、アルシ
ン、ホスフィン等いわゆる特殊材料ガス（強い
毒性と可燃性を持つガス）が使用されています。これらのガスは微量濃度（数 ppm～数十 ppm）のガス漏洩も許されません。
半導体/液晶パネル工場には ppm オーダーのガス漏洩を検知できる定電位電解式センサを搭載した当社のガス検知器が数百～数千個設置されており、ガス漏洩から作業者を守ります。

3.鉄鋼業界製鉄の各工程で発生する副生ガス（コークス
ガス、高炉ガス、転炉ガス）は水素や一酸化炭素を大量に含んでおり、製鉄所ではこれらのガスを発電用の燃料として使用し再利用しています。
当社のポータブルガス検知器は製鉄所内での爆発事故や中毒事故の危険から作業者を守ります。

4.石油精製 / 石油化学産業
石油精製・石油化学産業では、製造工程において多種多様の可燃性ガス・毒性ガスが使用されています。設備や配管からの可燃性ガ
ス・毒性ガスの漏洩検知やプロセス管理・作業
環境測定などの用途で当社の定置式ガス検知器やポータブルガス検知器が使用されています。
また、近年は工場境界線の毒性ガス管理用として、定置式毒性ガス環境モニターも多く使用されています。

- 42 -

6. ガス検知器の用途・設置場所(Typical areas that require gas detection)
6-1. ガス検知器の市場

5.火山･温泉
火口や噴出口付近や温泉が湧き出す地域では火山性ガスが発生します。この火山性ガス中には吸引すると人体に影響を与える毒性の二酸化硫黄や硫化水素が含まれています。これらの濃度は火山活動等により絶えず変化をしています。
当社の定置型の二酸化硫黄計や硫化水素計
は、ガス濃度を常時監視し、作業者や観光客を中毒事故から守ります。

6.食品業界食品業界では食品の酸化を防ぐ為、パッケー
ジに窒素や二酸化炭素を使用しています。これらのガスは窒息性ガスですので、酸素欠乏症から作業者を守るため食品工場に当社の酸素検知器が設置されています。

7.建設業界地下トンネル建造などの地下掘削作業やマンホール内での作業は、地層に含まれているバクテリアにより酸素が消費され酸欠状態となったり、硫化水素が発生したりします。また地中からメタン等が発生する場合があります。酸素欠乏症や硫化水素中毒事故および爆発事故の危険から作業者を守るため、当社のポータブル式の酸素検知器や硫化水素検知器が使用されています。

- 43 -

6. ガス検知器の用途・設置場所(Typical areas that require gas detection)
6-1. ガス検知器の市場

8.消防 / レスキュー
火災現場や災害現場では可燃性ガスによる爆発の危険や、酸素欠乏、不完全燃焼による一酸化炭素中毒、さらには硫化水素などの毒性ガスの危険等、様々な危険に隣り合わせです。
どんな危険なガスが発生しているかわからない現場に対応するため、4~６種類のガスが同時に測定できる当社のマルチガスモニターが使用されています。

9.海運 / 造船
原油、LNG、LPG を大量輸送する船舶は、貨物タンクからの可燃性ガス漏洩の危険性があります。当社の定置式ガス検知器はガス漏洩を監視及び早期検知し、漏洩による爆発や海上汚染を防止します。
また、建造作業中は当社のポータブルガス検知器を携帯することで危険な酸欠事故・毒性
ガス中毒事故から作業者を守ります。

10.航空 / 宇宙
ロケット燃料には爆発性の高い可燃性ガスの水素や人体に影響を及ぼす毒性ガスのヒドラジンが使用されており、安全のためには監視が必要です。
ロケット燃料を扱う現場のように爆発の危険が高い場所の安全確保には、防爆型のガス検知器が使用されています。

- 44 -

7. ガス検知技術(Gas detection technologies)
7-1. 理研計器のガスセンサ技術

7
7.

ガス検知技術(Gas detection
technologies)

7-1. 理研計器のガスセンサ技術
理研計器では幅広い業界での多様な環境、ガス種に対応するため、数多くのガスセンサ技術をもっています。本書では産業で代表的に使用される下記 13 種類についてご紹介します。
1.

接触燃焼式（Catalytic Combustion Method）

2.

ニューセラミック式

3.

半導体式

4.

熱線型半導体式

5.

熱伝導式

6.

定電位電解式

7.

隔膜分離型定電位電解式

8.

隔膜ガルバニ電池式

9.

非分散型赤外線式

10.

光波干渉式

11.

検知テープ式

12.

光イオン式

13

熱粒子化式

- 45 -

7. ガス検知技術(Gas detection technologies)
7-2. 接触燃焼式(Catalytic Combustion Method)

7-2. 接触燃焼式(Catalytic Combustion Method)- 46 -

7. ガス検知技術(Gas detection technologies)
7-3. ニューセラミック式(Catalytic Combustion Method)

7-3. ニューセラミック式(Catalytic Combustion Method)- 47 -

7. ガス検知技術(Gas detection technologies)
7-4. 半導体式(Semi-Conductor Method)

7-4. 半導体式(Semi-Conductor Method)- 48 -

7. ガス検知技術(Gas detection technologies)
7-5. 熱線型半導体式(Hot Wire Type Semi-Conductor Method )7-5. 熱線型半導体式(Hot Wire Type Semi-Conductor Method )- 49 -

7. ガス検知技術(Gas detection technologies)
7-6. 熱伝導式(Thermal Conductivity Method)

7-6. 熱伝導式(Thermal Conductivity Method)- 50 -

7. ガス検知技術(Gas detection technologies)
7-7. 定電位電解式(Potentiostatic Electrolysis Method)

7-7. 定電位電解式(Potentiostatic Electrolysis Method)- 51 -

7. ガス検知技術(Gas detection technologies)
7-8. 隔膜分離型定電位電解式(Membrane-Separated Electrode Method )7-8. 隔膜分離型定電位電解式(Membrane-Separated Electrode
Method )- 52 -

7. ガス検知技術(Gas detection technologies)
7-9. 隔膜ガルバニ電池式(Membrane Type Galvanic Cell Method )7-9. 隔膜ガルバニ電池式(Membrane Type Galvanic Cell
Method )- 53 -

7. ガス検知技術(Gas detection technologies)
7-10. 非分散型赤外線式(Non-Dispersive Infrared Method)

7-10. 非分散型赤外線式(Non-Dispersive Infrared Method)- 54 -

7. ガス検知技術(Gas detection technologies)
7-11. 光波干渉式(Non-Dispersive Infrared Method)

7-11. 光波干渉式(Non-Dispersive Infrared Method)- 55 -

7. ガス検知技術(Gas detection technologies)
7-12. 検知テープ式(Chemical Tape Method)

7-12. 検知テープ式(Chemical Tape Method)- 56 -

7. ガス検知技術(Gas detection technologies)
7-13.光イオン化式センサ(Photo-Ionization Detector)

7-13.光イオン化式センサ(Photo-Ionization Detector)- 57 -

7. ガス検知技術(Gas detection technologies)
7-14.熱粒子化式センサ(Pyrolysis-Particle Detection Method)

7-14.熱粒子化式センサ(Pyrolysis-Particle Detection Method)- 58 -

8. ガス検知器の選定(Selecting the proper measurement method)
8-1. ガス検知器の選定

88.

ガス検知器の選定(Selecting theproper measurement
method)8-1. ガス検知器の選定
ガスを検知する原理は様々ありますが、危険なガスを検知するためには測定対象ガス種や検知
濃度、環境、用途に応じた正しい検知原理を選択する必要があります。
1. 測定ガスは何か？
① 可燃性ガス(爆発防止用)
主に接触燃焼式、ニューセラミック式、半導体式、赤外線式、光波干渉式の 4 種類の検知方法があります。接触燃焼式センサは主に％LEL レンジ、ニューセラミック式は 1
万～数千 ppm の検知に、半導体式は数千～数十 ppm の測定レンジの場合に使用します。赤外線式、光波干渉式の可燃性センサは主に％LEL～vol％の濃度のガスを測定します。赤外線式センサ、光波干渉式センサは化学反応を伴わない物理センサのため、接触燃焼式センサや半導体式センサに対して被毒するような物質(ハロゲ
ン化物、硫黄化合物、シリコーン)の存在下でも安定したガス検知器が可能です。
② 毒性ガス(中毒防止用)
毒性ガスは主に数百 ppm～数 ppb でも検出できる高い感度のセンサが必要になります。半導体式、定電位電解式、熱粒子化式、テープ式、PID 式の検知方法があります。一般的には警報設定値：許容濃度が検知可能なレンジの検知原理を選定し- 59 -

8. ガス検知器の選定(Selecting the proper measurement method)
8-1. ガス検知器の選定ます。半導体式センサは数十 ppm～数千 ppm、定電位電解式は数十 ppm から
数 10ppb のオーダーのガスを検出します。熱粒子化式センサは TEOS(テトラエトキシシラン)等の半導体材料ガスの内、有機金属化合物の検知に特化したセンサ原理です。テープ式ガス検知器は数 ppb オーダーの極低濃度のガス検知や、干渉ガス影響を受けにくい特長があるため、他原理では干渉してしまう環境でも適用できる検知原理になります。

③ 酸素(酸素欠乏症、過剰酸素防止)
酸素を検知する原理は隔膜ガルバニ電池式、定電位電解式の２つの原理があります。
世の中のガス検知器で最も多く使用されているのは長期安定性が高く、干渉等の影響が少ない、隔膜ガルバに電池式の酸素センサですが、原理上、鉛(Pb)を使用する
ため、将来的に RoHS 規制対象になりうるセンサです(2023 年時点は適用除外製品に含まれます)。そのための対策として、鉛フリーの定電位電解式センサがラインナップ
されています。

2. 定置型またはポータブル型？
作業者が体に身に着けたり、または携帯して使用する場合にはポータブル型のガス検知器を選定してください。ガスの漏洩を固定した位置で監視する場合には定置式のガス検知器を選定して下さい。

3. 拡散式または吸引式
ガス検知器には拡散式と吸引式の 2 つのタイプの検知方式があります。吸引式のガス検知
器は機器にポンプを内蔵しており、漏洩しやすい箇所(ライン中や筐体内部等)からセンサ部までガスを吸引して検知するタイプのガス検知器です。拡散式のガス検知器は漂ってくるガスがセンサに到達した場合に検知する受動的なタイプのガス検知器になります。

4. マルチガス検知またはシングルガス検知
ポータブルガス検知器の中には単成分のガスを検知するガス検知器の他に、複数のガスを同時に検知できるガス検知器があります。マルチガス検知器の最も基本的な組み合わせは
可燃性ガス、毒性ガス(H2S、CO)、酸素の 4 つです。製品によっては他のガスセンサもライ
ンナップしており、検知したいガス種を自由に組み合わせて選択ができます。

- 60 -

9. 定置式ガス検知器(Fixed Gas Detectors)
9-1. 防爆型ガス検知器

99.

定置式ガス検知器(Fixed Gas
Detectors)

9-1. 防爆型ガス検知器
耐圧防爆型ガス検知器仕様- 61 -

9. 定置式ガス検知器(Fixed Gas Detectors)
9-1. 防爆型ガス検知器

耐圧防爆型ガス検知器(高温炉内対応型)※GD-A2400 は表示部・検知部セパレートタイプ

耐圧防爆型ガス検知器(表示部・検知部セパレートタイプ)- 62 -

9. 定置式ガス検知器(Fixed Gas Detectors)
9-1. 防爆型ガス検知器

耐圧防爆型ガス検知器(ポンプ内蔵型)
表示部・検知部セパレートタイプ

SD-D58
GD-A80GD-D58GD-1OX
GD-K88Ai/Di
GD-F88Ai/Di
GD-F3A-A/GD-F4A-A
D58 シリーズ

表示部付 4-20mA 伝送タイプ

耐圧防爆型ポンプ(センサ無)- 63 -

9. 定置式ガス検知器(Fixed Gas Detectors)
9-1. 防爆型ガス検知器

防爆型ガス検知器(高精度・高速応答・メンテフリー)- 64 -

9. 定置式ガス検知器(Fixed Gas Detectors)
9-1. 防爆型ガス検知器- 65 -

9. 定置式ガス検知器(Fixed Gas Detectors)
9-1. 防爆型ガス検知器- 66 -

9. 定置式ガス検知器(Fixed Gas Detectors)
9-1. 防爆型ガス検知器
本質安全防爆型毒性ガス検知器(ZONE 0 で使用可能)本質安全防爆型酸素検知器(ZONE 0 で使用可能)本質安全防爆型酸素検知器(ZONE 0 で使用可能, 表示部・検知部セパレートタイプ)- 67 -

9. 定置式ガス検知器(Fixed Gas Detectors)
9-2. 非防爆型ガス検知器

9-2. 非防爆型ガス検知器
半導体材料ガス検知器(可燃性・毒性・酸素)
主な検知対象ガス※NF3,TEOS の場合は別途熱分解器ユニットが必要になります。

半導体材料ガス検知器(C5F8/C4F6/COS 専用)- 68 -

9. 定置式ガス検知器(Fixed Gas Detectors)
9-2. 非防爆型ガス検知器
テープ式ガス検知器(高感度・高選択制型)
汎用

C5F8,C4F6 専用

高感度型- 69 -

9. 定置式ガス検知器(Fixed Gas Detectors)
9-2. 非防爆型ガス検知器- 70 -

9. 定置式ガス検知器(Fixed Gas Detectors)
9-2. 非防爆型ガス検知器- 71 -

9. 定置式ガス検知器(Fixed Gas Detectors)
9-2. 非防爆型ガス検知器
小型ガスモニター(簡易設置型)

フロン・PFC ガス検知器- 72 -

9. 定置式ガス検知器(Fixed Gas Detectors)
9-3. 指示警報部
溶剤ガスモニター(高精度・高速応答・メンテフリー)

9-3. 指示警報部
多点式指示警報部
接続例- 73 -

9. 定置式ガス検知器(Fixed Gas Detectors)
9-3. 指示警報部
多点式指示警報部(保安電源搭載型)
GP-148

一点式指示警報部 RM-6000 シリーズ
接続例

- 74 -

10. ポータブルガス検知器(Portable Gas detectors)

1010.

ポータブルガス検知器(Portable
Gas detectors)

10-1. 複合型ガス検知器(Multi-sensor)
ポケッタブルマルチガスモニター- 75 -

10. ポータブルガス検知器(Portable Gas detectors)- 76 -

10. ポータブルガス検知器(Portable Gas detectors)- 77 -

10. ポータブルガス検知器(Portable Gas detectors)- 78 -

10. ポータブルガス検知器(Portable Gas detectors)- 79 -

10. ポータブルガス検知器(Portable Gas detectors)- 80 -

10. ポータブルガス検知器(Portable Gas detectors)- 81 -

10. ポータブルガス検知器(Portable Gas detectors)

10-2. 単成分ガス検知器(Single-sensor)- 82 -

10. ポータブルガス検知器(Portable Gas detectors)

可燃性単成分ガスモニター- 83 -

10. ポータブルガス検知器(Portable Gas detectors)- 84 -

10. ポータブルガス検知器(Portable Gas detectors)- 85 -

10. ポータブルガス検知器(Portable Gas detectors)- 86 -

11. 代表的なアプリケーション(Applications for gas detectors)

1111.

代表的なアプリケーション(Applications for gas
detectors)
11-1. 自動車市場のアプリケーション- 87 -

11. 代表的なアプリケーション(Applications for gas detectors)- 88 -

11-1. 自動車市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 89 -

11-1. 自動車市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 90 -

11-1. 自動車市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 91 -

11-1. 自動車市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 92 -

11-1. 自動車市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 93 -

11-1. 自動車市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)

11-3. 鉄鋼市場のアプリケーション- 94 -

11-3. 鉄鋼市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 95 -

11-3. 鉄鋼市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 96 -

11-3. 鉄鋼市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 97 -

11-3. 鉄鋼市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)

11-4. 燻蒸市場のアプリケーション- 98 -

11-4. 燻蒸市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 99 -

11-4. 燻蒸市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 100 -

11-4. 燻蒸市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)

11-5. 製薬市場のアプリケーション- 101 -

11-5. 製薬市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 102 -

11-5. 製薬市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 103 -

11-5. 製薬市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 104 -

11-5. 製薬市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 105 -

11-5. 製薬市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)

11-6. 食品市場のアプリケーション- 106 -

11-6. 食品市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 107 -

11-6. 食品市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 108 -

11-6. 食品市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 109 -

11-6. 食品市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 110 -

11-6. 食品市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 111 -

11-6. 食品市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 112 -

11-6. 食品市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 113 -

11-6. 食品市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)

11-7. 大学・研究施設のアプリケーション- 114 -

11-7. 大学・研究施設のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 115 -

11-7. 大学・研究施設のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 116 -

11-7. 大学・研究施設のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 117 -

11-7. 大学・研究施設のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 118 -

11-7. 大学・研究施設のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 119 -

11-7. 大学・研究施設のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)

11-8. ガス市場のアプリケーション- 120 -

11-8. ガス市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 121 -

11-8. ガス市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 122 -

11-8. ガス市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 123 -

11-8. ガス市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)

11-9. 電力市場のアプリケーション- 124 -

11-9. 電力市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 125 -

11-9. 電力市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 126 -

11-9. 電力市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 127 -

11-9. 電力市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 128 -

11-9. 電力市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 129 -

11-9. 電力市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 130 -

11-9. 電力市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 131 -

11-9. 電力市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 132 -

11-9. 電力市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 133 -

11-9. 電力市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)

11-10. 製紙市場のアプリケーション- 134 -

11-10. 製紙市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 135 -

11-10. 製紙市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 136 -

11-10. 製紙市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 137 -

11-10. 製紙市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 138 -

11-10. 製紙市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)

11-11. 石油・石油化学市場のアプリケーション

11-11. 石油・石油化学市場のアプリケーション- 139 -

11. 代表的なアプリケーション(Applications for gas detectors)- 140 -

11-11. 石油・石油化学市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 141 -

11-11. 石油・石油化学市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 142 -

11-11. 石油・石油化学市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 143 -

11-11. 石油・石油化学市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 144 -

11-11. 石油・石油化学市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)

11-12. 上下水道市場のアプリケーション- 145 -

11-12. 上下水道市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 146 -

11-12. 上下水道市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 147 -

11-12. 上下水道市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 148 -

11-12. 上下水道市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 149 -

11-12. 上下水道市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 150 -

11-12. 上下水道市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 151 -

11-12. 上下水道市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 152 -

11-12. 上下水道市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)

11-13. 電子・デバイス市場のアプリケーション

11-13. 電子・デバイス市場のアプリケーション- 153 -

11. 代表的なアプリケーション(Applications for gas detectors)- 154 -

11-13. 電子・デバイス市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 155 -

11-13. 電子・デバイス市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 156 -

11-13. 電子・デバイス市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 157 -

11-13. 電子・デバイス市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 158 -

11-13. 電子・デバイス市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 159 -

11-13. 電子・デバイス市場のアプリケーション

11. 代表的なアプリケーション(Applications for gas detectors)- 160 -

11-13. 電子・デバイス市場のアプリケーション

12. 国際規格および認証（Global standards and approvals）

12-1. IEC 規格1212.

国際規格および認証（Global

standards and approvals）
国際規格は国際的に取り決められた規格で、一般的に国際的な取引をスムーズにするため、製品やサービスに関して世界中で同じ品質、同じレベルのものを提供できるようにすることを目的に制定されています。代表的なものとして電気分野を専門に取り扱う IEC 規格と、非電気分野を取り扱う ISO
規格があります。

12-1. IEC 規格
IEC 規格とは、IEC(International Electrotechnical Commission：国際電気標準会議)
によって制定される、電気・電子技術および関連技術に関する国際規格です。
IEC は 1906 年に日本を含む 13 ヵ国の代表により創設され、現在 88 ヵ国(2022 年 12 月)が加盟しています。
1 ヵ国 1 機関のみが国内委員会(National Committee)として認められ、日本では日本工業標準調査会(Japanese Industrial Standards Committee：JISC)が加盟しています。
加盟国は各国家規格について IEC 規格と整合性をとることが求められ、日本でも近年ガス検知器に関連する規格の IEC 規格との整合化が進められています。

12-2. IECEx
IEC は IEC 規格の制定だけでなく、IEC 規格に適合した製品の品質と安全性を保証する適合性評価制度を提供しています。
その一つに IECEx(IEC System for Certification to Standards Relating to Equipment
for Use in Explosive Atmospheres：IEC 防爆電気機器規格適合試験制度)があります。
IECEx は爆発性雰囲気内で使用される電気機器の一定の安全性を確保しつつ、従来各国ごとに行われる試験を共通化することで手続きを簡略化し、製品の国際的な流通を促進することを目的としています。
現在、IECEx 適合品をそのまま自国の防爆制度適合品として受け入れている国はオーストラリア、
ニュージーランドなど数カ国のみですが、日本を含む各国で IEC 規格と整合化する動きが進んでいます。
- 161 -

12. 国際規格および認証（Global standards and approvals）

12-3. 理研計器製品に関わる IEC 規格

12-3. 理研計器製品に関わる IEC 規格
・ IEC 60079-0 Explosive atmospheres - Part 0: Equipment - General requirements(爆発性雰囲気－第 0 部：機器－一般要求事項)
爆発性雰囲気での使用を意図した防爆機器及び防爆部品の構造及び試験及びマーキングの一
般的な要求事項を規定しています。IECEx で求められる規格です。
・ IEC 60079-1 Electrical apparatus for explosive gas atmospheres - Part 1:
Flameproof enclosures 'd'(爆発性雰囲気で使用する電気機械器具－第 1 部：耐圧防爆容器'd' )爆発性雰囲気での使用を意図した耐圧防爆構造機器の構造及び試験に関して規定しています。IECEx で求められる規格です。
・ IEC 60079-11 Electrical apparatus for explosive gas atmospheres - Part 11:
Intrinsic safety 'i'(爆発性雰囲気で使用する電気機械器具－第１１部：本質安全防爆構造"ｉ" )爆発性雰囲気での使用を意図した本質安全防爆構造機器の構造及び試験に関して規定しています。IECEx で求められる規格です。
・ IEC 61508 Functional safety of electrical/electronic /programmable electronic
safety-related systems(電気・電子・プログラマブル電子安全関連系の機能安全)
プロセス産業における電気・電子・プログラマブル電子(E/E/PE)の機能安全について規定し、
Part1～7 で構成されます。
機能安全とは、安全を確保する機能を導入して、許容できるレベルの安全を確保することです。
IEC61508 では安全関連系の構想段階から設計開発、保守・廃棄にいたるまで要求事項が定められ、リスクを分析し軽減することが求められます。
安全度合いは SIL(Safety Integrity Level：安全度水準)というシステムの安全性能を表す尺度で表されます。SIL は 1～4 の 4 段階で定められ、SIL4 が最高水準です。
- Part 1: General requirements (第 1 部：一般要求事項)
電気・電子・プログラマブル電子(E/E/PE)系について、安全機能の履行に使用する場合に必要となる考え方について規定しています。
- Part 2: Requirements for electrical/electronic/programmable electronic
safety-related systems (第 2 部：電気・電子・プログラマブル電子安全関連系の要求事項 )IEC 61508-1 に従って開発された E/E/PE システム安全要求仕様を E/E/PE システムの設計要求仕様に改良する方法を規定しています。
- Part 3: Software requirements (第 3 部：ソフトウェア要求事項 )安全関連システムの一部を形成する、または安全関連システムを開発するために使用されるソフト
ウェアに関して規定しています。
- Part 4: Definitions and abbreviations (第 4 部：用語の定義及び略語 )機能安全に関する用語の定義をしています。
- Part 5: Examples of methods for the determination of safety integrity levels（第 5 部：安全度水準決定方法の事例）- 162 -

12. 国際規格および認証（Global standards and approvals）

12-4. ISO 規格

安全関連系の信頼度、すなわち安全水準（SIL）を解析で目留める事例を紹介しています。
- Part 6: Guidelines on the application of IEC 61508-2 and IEC 61508-3（第 6部：第 2 部、第 3 部の適用指針）
プログラマブル電子系の信頼度を、アーキテクチャ校正を基に算出している事例を紹介しています。
- Part 7: Overview of techniques and measures（第 7 部：技術および手法の概観）
第 2 部、第 3 部に関係する安全技術・手法の説明と紹介をしています

12-4. ISO 規格
ISO 規格とは、ISO(International Organization for Standardization：国際標準化機構)によって制定される国際規格です。
電気・電子技術および関連技術を除く全産業分野(鉱工業、農業、医薬品等)を対象とし、製品そのものに対する要求だけでなく、組織のあり方などについても幅広く規定しています。ISO は 1947
年に設立され、現在 167 ヵ国(2022 年 12 月)が加盟しています。IEC 同様、ISO への加盟は 1 ヵ
国 1 機関のみと定められ、日本では日本工業標準調査会(JISC)が加盟しています。また、加盟国
は各国家規格について ISO 規格と整合性をとることが求められます。

12-5. 理研計器製品に関わる ISO 規格
ISO 9001 Quality management system - Requirements(品質マネジメントシステム－要求事項)
一定水準の製品・サービスの提供や顧客満足の向上を目的に、組織のマネジメントに関する要求事項を規定しています。
ISO 14001 Environmental management systems - Requirements with guidance
for use(環境マネジメントシステム－要求事項及び利用の手引き)
環境保護を目的に、組織のマネジメントに関する要求事項を規定しています。

- 163 -

13. 地域規格(Regional standards)

13-1. 欧州指令と CE マーキング

13
13.

地

域

規

格

(Regional

standards)
地域規格は、ある特定の地域内で限定的に利用される規格です。代表的なものとして、EU 加盟国
内で適用される EN 規格があります。地域規格は国際規格と協調することにより、内容の統一化が図られる傾向にあります。

13-1. 欧州指令と CE マーキング
欧州には、欧州域内での製品の自由な流通を促進するため、CE マーキングという制度があります。
対象国は欧州連合(European Union: EU)加盟国に加え、欧州自由貿易連合(European
Free Trade Association：EFTA)加盟国およびトルコで、これらの国で製品を販売するには、欧州委員会(European Commission：EC)が制定した欧州共通の法令である欧州指令に適合することが義務付けられます。 そのため、欧州指令が適用される製品については、その要求事項を満たし、かつ満たしていることを示す CE マークを貼付する必要があります。 欧州指令は幅広い分野の製品を対象に制定され、ガス検知器のような電気機器に対する EMC 指令、防爆機器に対する ATEX
指令など、特に安全性が求められる製品を中心に規定されています。 製品によっては複数の指令に該当する場合があり、例えば防爆のガス検知器の場合、ATEX 指令、EMC 指令、RoHS 指令に適合する必要があります。
しかし、CE マークはどの製品に対しても同じデザインのため、製品がどの指令に適合しているのかは CE
マークだけを見ても分かりません。適合している指令等の詳細については、各製造業者が発行する「自己宣言書(Declaration of Conformity：DoC)」にて確認することができます。

- 164 -

13. 地域規格(Regional standards)

13-2. 理研製品に関わる欧州指令13-2. 理研製品に関わる欧州指令
・

防爆指令(2014/34/EU)- Equipment for explosive atmospheres (ATEX)
爆発の恐れのある環境で使用する機器に適用されます。そのような環境下で使用する機器は、爆発を誘引しないよう防爆構造である必要があります。

・EMC 指令(2014/30/EU)- Electromagnetic Compatibility (EMC)
電磁波を発生する、または外部からの電磁波の影響を受ける可能性がある機器に適用されます。
EMC(Electromagnetic Compatibility)とは「電磁両立性」を意味し、機器は他の機器に影響を与えないだけでなく、他の機器からの影響を受けないことも求められます。

・RoHS 指令(2011/65/EU)- Restriction of the use of certain hazardous
substances (RoHS)
電気・電子機器に適用され、機器は鉛や水銀などの物質有害の含有量が基準値以下でなくてはなりません。RoHS 指令は 2 回改正され、現在は 10 物質が対象物質となっています。

・

低電圧指令(2014/35/EU)- Low voltage (LVD)
AC50 ～ 1000V、DC75 ～ 1500V で動作する機器に適用されます。電気機器の使用による感電や火災など危険を規制しています。

・

無線機器指令(2014/53/EU)- Radio equipment
無線機能を搭載した機器に適用されます。EMC 指令および LVD 指令の要求事項への適合や無線スペクトルという技術の効果的な使用が求められます。

・MED 指令(2014/90/EU )- Marine equipment directive
舶用機器に適用されます。ガス検知器については可燃性ガスおよび酸素センサの性能や船舶に特化した環境試験への適合が求められます。

13-3. 欧州規格(EN 規格)
欧州規格(European Norm：EN)とは、加盟国間の貿易を円滑にし、EU 内で流通する製品が一定水準の基準を満たすことを目的に制定される欧州の統一規格です。 電気分野に関する規格は欧州電気標準化委員会(CENELEC)が、非電気分野に関する規格は欧州標準化委員会(CEN)が制定しています。 規格単体では強制力を伴わない任意の規格ですが、各指令で整合規格として参照された場合は必ず満たさなければなりません。 なお、欧州加盟国は、欧州規格を自国の国家規格として採用することが義務付けられています。

- 165 -

13. 地域規格(Regional standards)

13-4. 理研計器製品に関する EN 規格

13-4. 理研計器製品に関する EN 規格＜EMC に関する EN 規格＞
・

EN 50270：Electromagnetic regatibility. Electrical apparatus for the detection
and measurement of combustible gases, toxic gases or oxygen
ガス検知器に対する EMC に関する要求事項を規定しています。

・EN 61326-1：Electrical equipment for measurement, control and laboratoryuse -- EMC requirements-Part 1: General requirements
交流 1000 V 以下もしくは直流 1500 V 以下の電源もしくは電池で動作する電気装置または測定対象の回路からの電源で動作する電気装置の EMC に関する要求事項を規定しています。
＜ソフトに関する EN 規格＞

・

EN 50271：Electrical apparatus for the detection and measurement of
combustible gases, toxic gases or oxygen. Requirements and tests forapparatus using software and/or digital technologies
ガス検知器のソフトウェアに関する要求事項を規定しています。
＜無線機能に関する EN 規格＞

・

EN 300 328：Wideband transmission systems; Data transmission equipment
operating in the 2,4 GHz band; Harmonised Standard for access to radio
spectrum
2.4GHz 帯における広帯域通信機の電波放射特性に関する要求事項を規定しています。

・EN 301 489-1：ElectroMagnetic Compatibility (EMC) standard for radio
equipment and services; Part 1: Common technical requirements;
Harmonised Standard for ElectroMagnetic Compatibility
無線機能を搭載した機器の EMC に関して、共通の技術的要求事項を規定しています。

・EN 301 489-17：ElectroMagnetic Compatibility (EMC) standard for radio
equipment and services; Part 17: Specific conditions for Broadband Data
Transmission Systems; Harmonised Standard for ElectroMagnetic
Compatibility
無線機能を搭載した機器の EMC に関して、広帯域データ伝送システムの特定条件について規定しています。
＜防爆に関する EN 規格＞

・

EN 50303：Group I, Category M1 equipment intended to remain functional inatmospheres endangered by firedamp and/or coal dust
火の粉や石炭の粉塵にさらされる鉱山や地上設備において使用する機器の設計や試験に関する要求事項を規定しています。

・EN IEC 60079-0：Explosive atmospheres. Equipment. General requirements
防爆機器に関する一般的な要求事項を規定しています。

- 166 -

13. 地域規格(Regional standards)

13-4. 理研計器製品に関する EN 規格・

EN 60079-1：Explosive atmospheres. Equipment protection by flameproof
enclosures "d"
耐圧防爆構造の機器に関する要求事項を規定しています。

・EN 60079-11：Explosive atmospheres. Equipment protection by intrinsic
safety "i"
本質安全防爆構造の機器に関する要求事項を規定しています。
＜RoHS に関する EN 規格＞

・

EN IEC 63000：Technical documentation for the assessment of electrical andelectronic products with respect to the restriction of hazardous substances
特定有害化学物質を含有していないこと証明するための手順や RoHS 指令への適合を宣言す
るために必要な技術文書について規定しています。
＜性能に関する規格＞

・EN 45544-1：Workplace atmospheres. Electrical apparatus used for the
direct detection and direct concentration measurement of toxic gases andvapours. General requirements and test methods
毒性ガス用のガス検知器に対する一般的な要求や試験方法について規定しています。

・EN 45544-2：Workplace atmospheres. Electrical apparatus used for the
direct detection and direct concentration measurement of toxic gases andvapours. Performance requirements for apparatus used for exposure
measurement
毒性ガス用のガス検知器に対する性能要求事項を規定しています。

・EN 45544-3：Workplace atmospheres. Electrical apparatus used for the
direct detection and direct concentration measurement of toxic gases andvapours. Performance requirements for apparatus used for general gas
detection
毒性ガス用のガス検知器に対する性能要求事項を規定しています。

・EN 45544-4：Workplace atmospheres. Electrical apparatus used for the
direct detection and direct concentration measurement of toxic gases andvapours. Guide for selection, installation, use and maintenance
毒性ガス用のガス検知器の選定、設置、使用およびメンテナンスに関するガイダンスです。

・EN 50104：Electrical equipment for the detection and measurement ofoxygen. Performance requirements and test methods
酸素用のガス検知器に対する性能要求事項や試験方法について規定しています。

・EN 60079-29-1：Explosive atmospheres. Gas detectors. Performancerequirements of detectors for flammable gases
可燃性ガス用のガス検知器の性能要求事項について規定しています。

- 167 -

13. 国家規格(National standards)

14-１. 日本

14
13.

国家規格(National standards)

国家規格は、特定の 1 国内で使われる規格で、日本では JIS がこれに属します。国家規格も地域規格同様、国際規格と協調することにより、内容の統一化が図られる傾向にあります。

14-１. 日本
14-1-1. JIS 規格
JIS とは、産業標準化法に基づき制定される日本の国家規格です。JIS は「日本産業規格※」を意味する"Japanese Industrial Standards"の頭文字をとった略称で、製品の規格や測定方法、サービスなどに関して幅広く規定しています。単体では強制力を伴わない任意の規格ですが、法律に引用されている場合など強制力を持つことがあります。製品供給者は、JIS への適合と合わせて、「自己適合宣言書」を発行し、どの規格に適合しているかを示します。理研計器も JIS 適合製品についてはホームページの各製品情報ページにて宣言書の PDF を提供しています。
JIS は主に関係団体などから構成される原案作成委員会にてまとめられた案を日本産業標準調査会(JISC)にて審議し、主務大臣によって制定されてきました。しかし、制定プロセスをより迅速に行うために、令和元年 7 月 1 日より、JISC を経ずに特定の民間の認定機関(認定産業標準作成機
関)から主務大臣に提案できるようになりました。認定機関としては日本規格協会(JSA)や日本鉄鋼連盟があります。各規格は JISC の Web サイトにて閲覧できます。
なお、規格の番号は、「JIS」に加えて部門を表すアルファベット 1 文字、数字 4～5 桁および発行年で表されます。
(例)JIS T8201：2010 → 酸素欠乏測定用酸素計(2010 年版)＜部門を表すアルファベット一覧＞
A(土木及び建築)
B(一般機械)C(電子機器及び電気機械)
D(自動車)
E(鉄道)- 168 -

13. 国家規格(National standards)

14-１. 日本

F(船舶)
G(鉄鋼)
H(非鉄金属)
K(化学)
L(繊維)
M(鉱山)
P(ﾊﾟﾙﾌﾟ及び紙)Q(管理ｼｽﾃﾑ)
R(窯業)
S(日用品)
T(医療安全用具)
W(航空)X(情報処理)Z(その他)※2019 年 7 月 1 日の法改正により、「日本工業規格」から「日本産業規格」に名称が変更されました。

14-1-2. 理研計器品関連する JIS 規格
・JIS T8201 - 酸素欠乏測定用酸素計 酸素欠乏の恐れがある環境で、酸素濃度測定や監視、または警報用として用いる酸素計に適用されます。酸素計性能や構造、試験方法について規定しています。
・JIS T8205 硫化水素計硫化水素の発生の恐れがある環境で、硫化水素濃度の測定や監視、または警報用として用いる硫化水素計に適用されます。硫化水素計の性能や構造、試験方法について規定しています。
・JIS T8206 - 可燃性ガス検知器
空気中に存在する可燃性のガスまたは蒸気の濃度を検知および測定するために用いる携帯式(ポータブル式)、可搬式、定置式ガス検知器に適用されます。可燃性ガス検知器の性能や構造、試験
方法について規定しています。なお、この規格は IEC 60079-29-1(2016 年版、第 2 版)をもととし、日本国内の実状に照らして一部内容を削除および変更して作成されています。
・JIS Q1000 - 適合性評価－製品規格への自己適合宣言指針 製品供給者が JIS への適合を翔栄する場合の自己関合宣言に対する一般要求事項について規定しています。

- 169 -

13. 国家規格(National standards)

14-2. 理研計器製品に関する EN 規格14-1-3. 防爆構造電気機械器具型式検定
防爆構造電気機械器具型式検定とは、日本の防爆検定で、日本国内の危険場所で防爆機器を設置、使用する場合は、労働安全衛生法(昭和 47 年法律第 57 号)第 44 条の２第１項に基づき登録型式検定機関による型式検定を受けなければならないとされています。一度型式検定に合格すると、製造者は有効期間中、同じ構造・仕様の機器の試験を受けずに何台でも製造し、販売することができます。有効期間は 3 年間で、3 年に一度更新が必要です。ただし、購入者は一度型式検定を受けて出荷された製品について、有効期間に限らず防爆製品として使用し続けることができます。検定合格品には合格品であることが分かるように検定合格年月、合格番号、合格証の交付を受けた者の名称が記載した型式検定合格標章が貼付されます。

14-1-4. JG 型式承認
JG 型式承認とは、国土交通省が日本船籍の船舶に搭載する法定舶用品に対して、所定の性能を満たしていることを確認する制度です。JG は「日本政府」を意味する"Japanese Government"の頭文字をとった略称です。対象製品について、JG は承認試験申請時に型式試験を実施し、承認された後は製品出荷時に日本舶用品検定協会(HK)が個別検定を実施します。JG 型式承認およびHK の個別検定をもって、持運び式(ポータブル式)ガス検知器は MSC.1Circ.1477 および SOLAS
XI-1 第 7 規則で持運び式ガス検知器の搭載が適用される国

14-2. 理研計器製品に関する EN 規格
CCC マーク認証とは、中国強制製品認証(China Compulsory Certification：CCC)制度という強制力を伴った認証制度です。
CCC マーク制度は中華人民共和国国家品質監督検験検疫総局(AQSIQ)(現 国家市場監督
管理総局(SAMR))および中国国家認証認定管理委員会(CNCA)によって指定される電気・電子製品などが対象で、中国国内で対象製品を販売するにはこの認証を受け、CCC マークを表示することが義務付けられます。
この認証制度の技術的基準は中国国家規格(GB 規格)で、IEC 規格や ISO 規格といった国際規格への整合化が進められています。

- 170 -

13. 国家規格(National standards)

14-3. 韓国―KCs マーク

14-3. 韓国―KCs マーク
KCs マーク認証とは、韓国産業安全公団(KOSHA)が管轄する強制力を伴った認証制度です。
この制度の対象となる製品は危険性の高い産業用機械や防爆機器、保護具などで、対象製品を韓国国内に輸出、販売、流通、設置、運転するにはこの認証を受け、KCs マークを表示することが義務付けられます。

14-4. ブラジル―INMETRO 認証
INMETRO 認証とはブラジル国家度量衡・規格・工業品質院(The National Institute ofMetrology, Standardization and Industrial Quality：INMETRO)による認証制度です。
INMETRO 認証には製造者が任意で認証を受ける自主認証と、製造および販売するために必ず求められる義務認証の 2 種類あります。理研計器製品が該当する防爆機器は後者の義務認証を取得する必要がある製品にあたり、その他にも医療機器、自動車関連部品など様々な製品が対象です。義務認証に該当する製品は、ブラジル国内で製造および販売するために認証を受けた上で製品登録し、適合していることを示すラベルを表示する必要があります。

14-5. アメリカ
3-5-1. FM 規格
FM 規格とは、FM(Factory Mutual Research Corporation)が制定する、主に工場設備
関連機器に対する製品安全規格です。FM は 1835 年に設立された火災保険会社で、独自に設けた基準である FM 規格に基づき試験を実施しています。FM は民間企業であり、FM 認証制度は強制力を伴った国家規格ではなく、任意の認証制度です。

3-5-2. UL 規格
UL 規格とは、UL LLC(Underwriters Laboratories Limited Liability Company、以下UL)が制定する、電気製品における火災や感電の危険性を規制する製品安全規格です。元はアメリカ保険業者安全試験所(Underwriters Laboratories Inc.)という火災保険業組合により設立された非営利試験機関で、UL 認証制度は強制力を伴った国家規格ではなく任意の認証制度です。しかし、米国の州によっては UL 認証を義務付けている場合もあり、広く採用されています。

- 171 -

13. 機器の保護等級(Ingress protection of enclosures)

14-5. アメリカ

15
13.

機 器 の 保 護 等 級 (Ingress

protection of enclosures)
筐体の保護等級
現在、液体や固形物が筐体に浸入した場合の保護等級を示すために、符号化された等級が幅広く使用されています。この分類は、構内の帯電部または可動部に接触した場合に、人を保護することも対象としています。ただし、これは、危険区域で使用される電子機器に関する保護等級を補足するものであって、代わりとなるものではないことに注意してください。
ヨーロッパでは、保護等級を示す符号は、”IP” という文字の後ろに、保護の度合いを示す 2 つの固有値が続きます。最初の数字は、内部の帯電部または可動部への接触に対する人の保護の度合いを表したもの、2 つめの数字は水が浸入した場合の筐体の保護を表したものです。たとえば、IP65 と示されている筐体は、帯電部または可動部への接触に対する完全な保護 (粉塵の防止)、および水噴霧または水流の浸入から保護されます。これは、コントローラーなどのガス検知機器に使用するのに適していま
すが、電子機器を適切に冷却するための注意も必要です。2 桁の IP 符号はイギリスでよく使用されている短い形式です。国際的な完全版では、IP の後に 2 桁ではなく 3 桁の数字が続きます (例: IP
653)。3 桁目は衝撃抵抗を示します。数字の意味は下の表のとおりです。

- 172 -

13. 機器の保護等級(Ingress protection of enclosures)

14-5. アメリカ

IP codes (IEC / EN 60529)
保護等級 (IP) のコード(IEC / EN 60529)
最初の数字

2 番目の数字

外来固形物に対する保護等級

IP

水の浸入に対する保護等級

無保護

0

0

無保護

直径 50mm 以上の大きさ

11

鉛直に落下する水に対して保護

直径 12.5mm 以上の大きさ

2215 度以内に傾斜しても鉛直に落下する
水滴に対して保護

直径 2.5mm 以上の大きさ

33

散水に対して保護

直径 1.0mm 以上の大きさ

44水の飛まつに対して保護

防じん形

5

5

噴流に対して保護

耐じん形

6

6

噴流に対して保護

7

水に浸しても影響がないように保護

8

潜水状態での使用に対して保護例 IP67 は防塵で、浸水の影響から保護されている。

北米では、NEMA システムを使用して筐体を評価します。下の表は、NEMA システムと IP 評価を比較したものです。
NEMA,UL,CSA
等級

おおよそ相当する IEC/IP
等級

1

IP20

屋内、内容物との接触から

2

IP22

屋内、限定的、落下する汚れと水

3

IP55

屋外での雨、みぞれ、風による粉塵、氷の被害からの保護

3R

IP24

屋外での雨、みぞれ、氷の被害から守る

4

IP66

4X

IP66

6

IP67

12

IP54

屋内・屋外を問わず、風による粉塵、水しぶき、ホースによる水や氷の飛散による損傷から保護
屋内・屋外を問わず、腐食、風によるほこり、雨、水しぶき、ホースによる水や氷による損傷から守る
屋内および屋外、ホースからの放水、水没時の浸水、氷による損
傷
内、ほこり、落下物、非腐食性液体の滴下から

13

IP54

屋内、ほこり、落下物、非腐食性液体の滴下から内容- 173 -

14. メンテナンスと保守について（Gas detection maintenance and ongoing care）

14-5. アメリカ

14
14.

メンテナンスと保守について（Gasdetection maintenance andongoing care）
ガス検知警報機は検知対象が極低濃度であったり、化学的な活性が高かったり、取扱が危険だったり、その上種類も多いことから、メンテナンスには専門的な知識や十分経験が必要な特殊な機器です。
また、使用環境が複雑多岐にわたり、過酷なことも多いことから機器の信頼性を維持するためには保守点検を行うことが極めて重要になります。
理研計器は世界各国に営業拠点、サービス拠点のネットワークをもっており、各国のサービス員にはそれぞれ定期的なメンテナンス講習を受講させ、修了証を発行することでメンテナンス業務のレベル向上とその維持を行っております。安心してガス検知警報機の保守点検をご用命いただきますようお願い申し上げます。
理研計器の営業・サービスネットワーク- 174 -