PT2cn-3510

带信号转换器气体检测仪SD-3 系列

使用说明书
技术手册

目录

目录
1.

产品概要 ............................................................................................................................................................... 5
1-1 前言................................................................................................................................................................ 5
1-2 使用目的和特点 ............................................................................................................................................. 7
1-2-1 型号及代码 .......................................................................................................................................... 7
1-2-2 各型号的检测原理和检测对象气体 ..................................................................................................... 8
1-2-3 本仪器的特点 ...................................................................................................................................... 9
1-3 可使用的区域和条件.................................................................................................................................... 10
1-4 危险、警告、注记的定义 ............................................................................................................................ 11
1-5 标准及防爆规格的确认方法.......................................................................................................................... 112.

安全上的重要事项............................................................................................................................................... 12
2-1 危险事项 ...................................................................................................................................................... 12
2-2 警告事项 ...................................................................................................................................................... 12
2-3 注意事项 ...................................................................................................................................................... 13
2-4 安全信息 ...................................................................................................................................................... 143.

产品组成 ............................................................................................................................................................. 31
3-1 主机及附件................................................................................................................................................... 31
3-1-1 主机(SD-3) ........................................................................................................................................ 31
3-1-2 主机(SD-3SC) ................................................................................................................................... 32
3-1-3 遥控传感器头(GD-3) ......................................................................................................................... 32
3-1-4 标准附件............................................................................................................................................ 33
3-1-5 特别附件（选配件） ......................................................................................................................... 34
3-2 各部名称与作用 ........................................................................................................................................... 37
3-3 框图.............................................................................................................................................................. 38

4.

安装方法 ............................................................................................................................................................. 40
4-1 有关安装的注意事项.................................................................................................................................... 40
4-1-1 有关安装的一般信息 ......................................................................................................................... 40
4-1-2 有关安装的限制信息 ......................................................................................................................... 41
4-2 系统设计上的注意事项 ................................................................................................................................ 45
4-2-1 使用稳定的电源................................................................................................................................. 45
4-2-2 防雷措施............................................................................................................................................ 45
4-3 确保维护空间 ............................................................................................................................................... 46
4-4 安装主机 ...................................................................................................................................................... 48
4-4-1 安装到垂直杆 .................................................................................................................................... 48
4-4-2 安装到墙面 ........................................................................................................................................ 49
4-4-3 安装保护罩 ........................................................................................................................................ 50
4-4-4 安装防晒罩 ........................................................................................................................................ 52
4-5 安装遥控传感器头(GD-3) ............................................................................................................................ 54
4-5-1 安装到墙面 ........................................................................................................................................ 54
4-5-2 安装到线槽 ........................................................................................................................................ 54
4-6 接地端子的连接 ........................................................................................................................................... 56
4-7 线缆的连接................................................................................................................................................... 57
4-7-1 主机的线缆和仪器的连接口 .............................................................................................................. 57
4-7-2 将线缆连接到主机 ............................................................................................................................. 59
4-7-3 将电源线缆和信号线缆连接到遥控传感器头(GD-3) ......................................................................... 62
4-7-4 连接主机(SD-3SC)和遥控传感器头(GD-3) ...................................................................................... 65
4-7-5 将线缆连接到端子排 ......................................................................................................................... 70
4-8 安装传感器................................................................................................................................................... 73
4-9 警报继电器的设定（选配件） ..................................................................................................................... 74
2 / 171目录

4-10 安装防溅罩................................................................................................................................................. 77
5.

使用方法 ............................................................................................................................................................. 79
5-1 启动准备 ...................................................................................................................................................... 79
5-2 启动.............................................................................................................................................................. 80
5-3 基本运行流程 ............................................................................................................................................... 81
5-4 操作方法 ...................................................................................................................................................... 82
5-5 用户模式 ...................................................................................................................................................... 83
5-5-1 用户模式的设定项目 ......................................................................................................................... 83
5-5-2 切换到用户模式................................................................................................................................. 84
5-6 进行零位校正 ............................................................................................................................................... 85
5-7 确认设定值................................................................................................................................................... 87
5-7-1 确认警报设定值................................................................................................................................. 87
5-8 结束.............................................................................................................................................................. 896.

警报动作 ............................................................................................................................................................. 90
6-1 气体警报运行 ............................................................................................................................................... 90
6-1-1 警报运行............................................................................................................................................ 90
6-1-2 接点运行（自动复位）...................................................................................................................... 91
6-1-3 接点运行（自我保持）...................................................................................................................... 92
6-1-4 发出警报时的应对措施...................................................................................................................... 95
6-2 故障警报运行 ............................................................................................................................................... 95
6-3 警告运行 ...................................................................................................................................................... 96
6-4 外部输出运行 ............................................................................................................................................... 97
6-4-1 测量气体非氧气时 ............................................................................................................................. 97
6-4-2 测量气体为氧气时 ........................................................................................................................... 101
6-5 发出正确警报所需的各种功能 ................................................................................................................... 104
6-5-1 零抑制功能 ...................................................................................................................................... 104
6-5-2 传感器自动识别功能 ....................................................................................................................... 105

7.

保养检查 ........................................................................................................................................................... 106
7-1 检查的频率与检查项目 .............................................................................................................................. 106
7-2 维护模式 .................................................................................................................................................... 108
7-2-1 维护模式的设定项目 ....................................................................................................................... 108
7-2-2 切换维护模式 .................................................................................................................................. 109
7-3 进行各种测试（测试模式）........................................................................................................................111
7-3-1 进行气体测试 ...................................................................................................................................111
7-3-2 进行警报测试 .................................................................................................................................. 112
7-3-3 进行故障测试 .................................................................................................................................. 114
7-3-4 进行 LED 测试................................................................................................................................. 115
7-3-5 进行存储器测试（RAM 测试）....................................................................................................... 116
7-4 进行气体校正 ............................................................................................................................................. 117
7-4-1 气体校正的准备............................................................................................................................... 117
7-4-2 进行零位校正 .................................................................................................................................. 124
7-4-3 进行间距校正 .................................................................................................................................. 125
7-5 进行环境设定 1 .......................................................................................................................................... 129
7-5-1 INHIBIT 设定 .................................................................................................................................... 129
7-5-2 警报点设定 ...................................................................................................................................... 130
7-5-3 日期时间设定 .................................................................................................................................. 131
7-6 进行环境设定 3 .......................................................................................................................................... 132
7-6-1 双量程外部输出设定 ..................................................................................................................... 133
7-7 部件的更换................................................................................................................................................. 134
7-7-1 传感器的更换 .................................................................................................................................. 134

3 / 171

目录

7-7-2 定期更换部件的更换 ....................................................................................................................... 135
8.

关于储存、迁移及报废 ..................................................................................................................................... 136
8-1 储存或长期不使用时的处理....................................................................................................................... 136
8-2 移机或重新使用时的处理 .......................................................................................................................... 138
8-3 产品的废弃................................................................................................................................................. 138

9.

故障排除 ........................................................................................................................................................... 140
9-1 仪器异常 .................................................................................................................................................... 140
9-2 指示值异常................................................................................................................................................. 141

10. 产品规格 ........................................................................................................................................................... 143
10-1 SD-3 规格 ................................................................................................................................................. 143
10-1-1 SD-3（无 EC 安全栅规格） .......................................................................................................... 143
10-1-2 SD-3（有 EC 安全栅规格） .......................................................................................................... 146
10-1-3 电气额定值(SD-3) ......................................................................................................................... 149
10-2 SD-3SC＋GD-3 规格 ............................................................................................................................... 150
10-2-1 SD-3SC＋GD-3（无 EC 安全栅规格） ........................................................................................ 150
10-2-2 SD-3SC＋GD-3（有 EC 安全栅规格） ........................................................................................ 153
10-2-3 电气额定值(SD-3SC+GD-3) ......................................................................................................... 156
10-3 各种原理的检测对象气体 ........................................................................................................................ 157
10-3-1 非分散型红外线吸收法（红外线式） ........................................................................................... 157
10-3-2 新型陶瓷式 .................................................................................................................................... 158
10-3-3 半导体式........................................................................................................................................ 159
10-3-4 热线型半导体式............................................................................................................................. 161
10-3-5 恒电位电解式 ................................................................................................................................ 163
11. 附录 ................................................................................................................................................................... 164
11-1 检测原理................................................................................................................................................... 164
11-1-1 非分散型红外线吸收法（红外线式）............................................................................................ 164
11-1-2 新型陶瓷式 .................................................................................................................................... 165
11-1-3 半导体式 ........................................................................................................................................ 166
11-1-4 热线型半导体式 ............................................................................................................................. 167
11-1-5 恒电位电解式................................................................................................................................. 168
11-2 术语的定义 ............................................................................................................................................... 169
产品保修 ........................................................................................................................................................... 170
传感器保修........................................................................................................................................................ 170
修订记录 ........................................................................................................................................................... 171

4 / 1711. 产品概要

1-1 前言1产品概要
1-1 前言感谢您购买 SD-3 系列带信号转换器气体检测仪（以下简称“本仪器”）。
本使用说明书讲解了本仪器的使用方法与规格。记载了正确使用本仪器的必要事项。
使用前请仔细阅读，在充分理解内容的基础上使用本仪器。
另外，请将本使用说明书保存在身边，以便使用本仪器时能随时查阅。
本说明书的内容可能因产品改良而发生变更，恕不另行通知。另外，禁止擅自复制或转载本说明书的全部或部分。
无论是否在保修期内，对因使用本仪器造成的任何事故及损害均不进行补偿。
请务必确认本书末尾的保修规定。
使用本仪器前，请确认购买的产品型号和本使用说明书与对象产品的型号是一致的。

＜本使用说明书的对象产品型号＞

主机

主机

遥控
传感器头

扩散式
SD-3RI
SD-3NC
SD-3GH
SD-3GHS
SD-3SP
SD-3EC
SD-3ECS
SD-3ECB
SD-3SCGD-3RI
GD-3NCGD-3GH
GD-3GHS
GD-3SP
GD-3ECGD-3ECS
GD-3ECB

抽吸式
SD-3DRI
SD-3DNC
SD-3DGH
SD-3DGHS
SD-3DSP
SD-3DEC
SD-3DECS
SD-3DECB

5 / 1711. 产品概要

1-1 前言＜关于本使用说明书＞
本使用说明书中，有些说明的内容因检测原理而异，分别用以下图标表示。
非分散型红外线吸收法（红外线式）

RI

新型陶瓷式

NC

半导体式

GH

热线型半导体式

SP

恒电位电解式EC

没有图标的操作方法及规格是所有检测原理通用的内容。
本使用说明书的显示例中，检测对象气体为可燃性气体（单位：%LEL）。

6 / 1711. 产品概要

1-2 使用目的和特点

1-2 使用目的和特点
本仪器是固定式防爆仪器，用于连续监视大气中的可燃性气体、毒性气体和氧气。
可检测大气中的可燃性气体、毒性气体和氧气，在达到或超过预设浓度时发出警报。
并不是根据检测结果保障生命、安全的仪器。

1-2-1 型号及代码
SD-3 ①

②

( ③

1

0

0

⑦

⑧ )

【遥控式／主机部分(SD-3SC)＋遥控传感器部分（GD-3 系列）】
SD-3SC ( ③
1
0
0
⑦
⑧ )
GD-3 ①
②
①

②

③

④

⑤

扩散/抽吸的选择无扩散式D
抽吸式（通过外部单元导入）传感器种类的选择
RI
红外线式
NC
新型陶瓷式
GH
半导体式
GHS
半导体式（仅 CS2 可选）SP热线型半导体式
EC
恒电位电解式（仅 CO/O2 可选）ECS恒电位电解式（仅 H2S 可选）
ECB恒电位电解式（CO/O2/H2S 以外可选）线缆连接口0
连接口 1＋连接口 2
1
连接口 1＋连接口 2＋连接口 4＋连接口 5防爆
1
ATEX/IECEx/UKEX
2―
3
Japan Ex
4
―
功能安全
0
无
1
―

⑥

⑦

⑧

性能检定
0
无
1
―
2
―
3
―
量程设定※
0
单量程
1
W 量程＋4-16（仅 NC 可选）
2
W 量程＋L4-20（仅 NC 可选）
3
W 量程＋H4-20（仅 NC 可选）
输出方式的选择
0
4-20mA with HART
1
4-20mA with HART＋接点(3c)2―※选择 W 量程时，无法使用 HART 通信。

连接口 3

※未使用

连接口 2

连接口 4

标准

选配件

连接口 1

连接口 5

标准

选配件
（例）线缆连接口

7 / 1711. 产品概要

1-2 使用目的和特点

1-2-2 各型号的检测原理和检测对象气体本仪器根据检测方式和检测原理有以下几种型号。
使用前请再次确认规格，根据目的正确进行气体检测。
＜各型号待检测气体列表＞
型号
主机

检测
方式

检测原理

检测对象气体
可燃性
毒性
氧气
气体
气体

SD-3RI

非分散型红外线吸收法（红外线式）〇

SD-3NC

新型陶瓷式〇

SD-3GH半导体式

〇

SD-3GHS

扩散

半导体式

备注

〇

〇
仅 CS2（二硫化碳）〇

SD-3SP热线型半导体式

SD-3EC恒电位电解式〇

SD-3ECS恒电位电解式〇仅 H2S（硫化氢）SD-3ECB恒电位电解式〇带 EC 安全栅※

SD-3DRI

非分散型红外线吸收法（红外线式）〇

SD-3DNC

新型陶瓷式〇

SD-3DGH半导体式

〇

SD-3DGHS

抽吸

〇

半导体式

〇
〇

〇

〇
仅 CS2（二硫化碳）〇

SD-3DSP热线型半导体式

SD-3DEC

恒电位电解式〇SD-3DECS

恒电位电解式〇仅 H2S（硫化氢）SD-3DECB

恒电位电解式〇带 EC 安全栅※※〇因检测对象气体而异。

8 / 171〇〇1. 产品概要

1-2 使用目的和特点＜各型号检测对象气体列表（遥控式）＞
型号 主机
型号
遥控
传感器头

SD-3SC

检测
方式

检测原理

检测对象气体
可燃性
毒性
氧气
气体
气体

GD-3RI

非分散型红外线吸收法（红外线式）〇

GD-3NC

新型陶瓷式〇

GD-3GH半导体式

〇

GD-3GHS

扩散

半导体式

备注

〇

〇
仅 CS2（二硫化碳）〇

GD-3SP热线型半导体式

GD-3EC恒电位电解式〇

GD-3ECS恒电位电解式〇仅 H2S（硫化氢）

GD-3ECB恒电位电解式〇带 EC 安全栅※

※

〇

〇
〇

因检测对象气体而异。

＜检测原理和传感器型号＞
传感器型号

检测原理

IRF

红外线式

NCF

新型陶瓷式

SGF

半导体式

SHF

热线型半导体式ESF

恒电位电解式

1-2-3 本仪器的特点＜标准规格＞
 本仪器由不锈钢外壳构成，适用于室内和室外两个用途。
 仪器运行需要接通电源 DC 24V(DC 18V - DC 30V)。
 气体浓度、状态和命令选择菜单显示在 7 段数码管 LED 上。
 检测的气体浓度值转换输出为 4 - 20mA 模拟输出（按照 NAMUR NE43 执行）或数字 HART 输出信号。
 可以通过密封接头或获认证的线缆防水接头连接到本公司的监视系统或可编程逻辑控制器(PLC)。

＜选配件规格＞
 可设定警报继电器，将本仪器作为独立单元进行操作。如果作为独立单元，不需要中央控制器即可操作仪器。※1※1

若要作为独立单元使用，需设定额外的警报信号。

9 / 1711. 产品概要

1-3 可使用的区域和条件

1-3 可使用的区域和条件
本仪器获批准使用于可能存在ⅡA、ⅡB 爆炸性气体群组或ⅡC 气体的场所。设计意图限用于本仪器使用温度范围
内的 Zone1 或 Zone2 危险区域。

10 / 1711. 产品概要

1-4 危险、警告、注记的定义

1-4 危险、警告、注记的定义
本使用说明书中，对于无视标识内容操作错误时产生的危害的程度是如下区分说明的。

危险

表示操作错误时“可能危及生命或对身体、财物造成重大损害”。

警告表示操作错误时“可能对身体、财物造成重大损害”。

注意

表示操作错误时“可能对身体、财物造成轻微损害”。

除此以外，对于操作上的建议，是如下标识并说明的。

注记

表示了解后有助于使用本仪器的事项。

1-5 标准及防爆规格的确认方法
根据不同的标准及防爆审定种类，本仪器的规格也会有所差异。使用前请确认所持有的产品规格。
此外，使用 CE/UKCA 标志规格时，请参照卷末的符合性声明（Declaration of Conformity）。产品规格可通过产品上如下图所示粘贴的铭牌进行确认。

< SD-3 >ATEX/IECEx/UKEX, CE/UKCA 标志规格的铭牌例CE 标志< GD-3 >UKCA 标志

11 / 1712. 安全上的重要事项

2-1 危险事项

2

安全上的重要事项
2-1 危险事项危险• 请勿在通电状态下打开盖子。如需打开盖子，请使用便携气体检测仪等确认周围没有气体。
• 请勿修理本仪器。
• 请勿修理、更换防爆接合面。容器或防爆接合面发现划痕、裂纹、变形等情况时，请立即停止使用，及时联系经销商或就近的本公司营业网点。

2-2 警告事项警告• 接通电源时，请务必先确认电压符合规定值后再为本仪器接通电源。
如果使用的电压不符合规定值，可能导致仪器损坏。另外，不稳定的电源可能导致误动作，因此请勿使用。
• 请勿切断本仪器的保护接地线或拆下保护接地端子的接线。
• 运行本仪器前，请确认保护功能没有缺陷。
如果判断保护接地等的保护功能有缺陷，请勿运行本仪器。
• 请切实做好保护接地后，再连接到外部控制电路。
• 使用周围空气进行零位校正时，请先确认周围是清洁的大气。
如果在存在杂质气体等的状态下进行零位校正，就无法进行正确的校正。另外，气体正在泄漏时无法正确检测，非常危险。
• 发出了气体警报时十分危险。请客户进行判断，在确保安全的基础上正确进行处理。
• 请勿长期将本仪器浸泡在有机溶剂中。构成火焰消除器的部件的材质是聚苯硫醚(PPS)树脂。

12 / 1712. 安全上的重要事项

2-3 注意事项

2-3 注意事项注意• 如果在本仪器或线缆附近使用收发器等发射电磁波，可能会影响本仪器工作。使用收发器等时，请在不影
响本仪器工作的地方使用。
• 如需重启电源，请间隔至少 10 秒。如果立即重启电源，仪器可能无法正常工作。
• 本仪器并非控制设备。禁止使用本仪器的外部输出功能来控制其他仪器。
• 请勿拆解或改造本仪器。拆解、改造本仪器将导致无法保证性能。
• 请勿在不解内容的情况下随意变更设定。某些情况下可能导致警报无法正常工作。除了本使用说明书中记
载的操作以外，请勿进行其他操作。
• 本仪器是安全仪器。为确保安全，请务必定期进行检查。如果不进行检查就持续使用，传感器的灵敏度劣化，就无法正常进行气体检测。

13 / 1712. 安全上的重要事项

2-4 安全信息

2-4 安全信息＜本仪器概要＞ 本仪器是耐压防爆结构的气体检测仪。
 根据检测原理，传感器部分成 3 种结构。采样方式有 2 种：扩散式和抽吸式。为抽吸式时，请安装使用抽吸帽。
 本仪器是固定式连续监视检测仪。连接到气体监视控制器、存储装置和可编程控制器，并将测量气体的指示值输出为 4 - 20mA 或 HART 信号。
 本仪器的输出规格有以下 3 个选项。
· 无警报继电器/无数字通信· 警报继电器（接点输出）×3（警报继电器与气体警报或故障联动工作。
）/无数字信号· 无警报继电器/数字通信（RS-485 输出） 通过使用 EC 安全栅，以电化学式传感器作为本质安全防爆结构。

＜检测原理和型号＞
 与各检测原理对应的型号如下所示。型号根据检测原理和采样方式进行分类。
检测原理（采样方式：扩散式）

型号

光传感器

非分散型红外线吸收法（红外线式）

SD-3RI

固体传感器

半导体式

SD-3GH
SD-3GHS电化学式传感器

新型陶瓷式

SD-3NC热线型半导体式

SD-3SP恒电位电解式

SD-3EC
SD-3ECS
SD-3ECB检测原理（采样方式：抽吸式）

型号

光传感器

非分散型红外线吸收法（红外线式）

SD-3DRI

固体传感器

半导体式

SD-3DGH
SD-3DGHS电化学式传感器

新型陶瓷式

SD-3DNC热线型半导体式

SD-3DSP恒电位电解式SD-3DEC
SD-3DECS
SD-3DECB

14 / 1712. 安全上的重要事项

2-4 安全信息 SD-3SC 与遥控传感器头(GD-3)组合时的型号如下所示。
型号
检测原理（采样方式：扩散式）
光传感器
固体传感器

非分散型红外线吸收法（红外线式）
半导体式

电化学式传感器

新型陶瓷式
热线型半导体式
恒电位电解式GD-3（遥控传感器头）GD-3RI
GD-3GH
GD-3GHS
GD-3NCGD-3SP
GD-3ECGD-3ECS
GD-3ECB

SD-3

SD-3SC

警告• SD-3EC、SD-3DEC 和 GD-3EC 的型号根据传感器上安装的火焰消除器的防爆结构和有无 EC 安全栅进行分类。
如果使用不同型号的火焰消除器，将无法保持原有的防爆性能。请勿更换成不同型号的火焰消除器。
• SD-3GH、SD-3DGH 和 GD-3GH 的型号根据传感器上安装的火焰消除器的防爆结构进行分类。
如果使用不同型号的火焰消除器，将无法保持原有的防爆性能。请勿更换成不同型号的火焰消除器。

15 / 1712. 安全上的重要事项

2-4 安全信息＜防爆性能数据＞
可以保持防爆性能的周围温度变化，取决于本仪器有无选配的避雷器。
请参阅以下各型号的性能数据。
＜无避雷器时＞
防爆
型号
结构

分组

分类

防爆等级

EPL

环境温度※1d

Ⅱ

2G

Ex db IIC T6/T5

Gb-50℃ ≤ Ta ≤ +60℃/+70℃

d

Ⅱ

2G

Ex db IIC T5/T4

Gb-50℃ ≤ Ta ≤ +44℃/+70℃

d

Ⅱ

2G

Ex db IIC T6/T4

Gb-50℃ ≤ Ta ≤ +47℃/+70℃

d

Ⅱ

2G

Ex db IIC T5/T4

Gb-50℃ ≤ Ta ≤ +44℃/+70℃

d

Ⅱ

2G

Ex db IIC T5/T4

Gb-50℃ ≤ Ta ≤ +55℃/+70℃

d

Ⅱ

2G

Ex db IIC T4

Gb-50℃ ≤ Ta ≤ +70℃

d

Ⅱ

2G

Ex db IIC T4

Gb-50℃ ≤ Ta ≤ +70℃

d+i

Ⅱ

2G

Ex db ia IIC T4

Gb-40℃ ≤ Ta ≤ +70℃

SD-3SC※2dⅡ

2G

Ex db IIC T6/T5/T4

Gb-50℃ ≤ Ta ≤ +47℃/+55℃+70℃

SD-3SC※3dⅡ

2G

Ex db IIC T5/T4

Gb-50℃ ≤ Ta ≤ +44℃/+70℃

SD-3SC

※4

d

Ⅱ

2G

Ex db IIC T6/T5

Gb-50℃ ≤ Ta ≤ +60℃/+70℃

SD-3SC

※5

d

Ⅱ

2G

Ex db IIC T4

Gb-40℃ ≤ Ta ≤ +70℃

SD-3RI
SD-3DRI
GD-3RI
SD-3GH
SD-3DGH
GD-3GH
SD-3GHS
SD-3DGHS
GD-3GHS
SD-3NC
SD-3DNC
GD-3NC
SD-3SP
SD-3DSP
GD-3SP
SD-3EC
SD-3DEC
GD-3EC
SD-3ECS
SD-3DECS
GD-3ECS
SD-3ECB
SD-3DECB
GD-3ECB※1 环境温度是指能够维持防爆性能的温度范围，并非满足产品性能的温度范围。关于使用温度范围，请参阅“10-3 各
种原理的检测对象气体”。
※2 与 GD-3GHS、GD-3SP、GD-3EC、GD-3ECS 组合时※3 与 GD-3GH、GD-3NC 组合时※4 与 GD-3RI 组合时※5 与 GD-3ECB 组合时

16 / 1712. 安全上的重要事项

＜有避雷器时＞
防爆型号
结构

2-4 安全信息

分组

分类

防爆等级

EPL

环境温度※1d

Ⅱ

2G

Ex db IIC T6/T5

Gb-40℃ ≤ Ta ≤ +60℃/+70℃

d

Ⅱ

2G

Ex db IIC T5/T4

Gb-40℃ ≤ Ta ≤ +44℃/+70℃

d

Ⅱ

2G

Ex db IIC T6/T4

Gb-40℃ ≤ Ta ≤ +47℃/+70℃

d

Ⅱ

2G

Ex db IIC T5/T4

Gb-40℃ ≤ Ta ≤ +44℃/+70℃

d

Ⅱ

2G

Ex db IIC T5/T4

Gb-40℃ ≤ Ta ≤ +55℃/+70℃

d

Ⅱ

2G

Ex db IIC T4

Gb-40℃ ≤ Ta ≤ +70℃

d

Ⅱ

2G

Ex db IIC T4

Gb-40℃ ≤ Ta ≤ +70℃

d+i

Ⅱ

2G

Ex db ia IIC T4

Gb-40℃ ≤ Ta ≤ +70℃

SD-3SC※2dⅡ

2G

Ex db IIC T6/T5/T4

Gb-40℃ ≤ Ta ≤ +47℃/+55℃+70℃

SD-3SC※3dⅡ

2G

Ex db IIC T5/T4

Gb-40℃ ≤ Ta ≤ +44℃/+70℃

SD-3SC

※4

d

Ⅱ

2G

Ex db IIC T6/T5

Gb-40℃ ≤ Ta ≤ +60℃/+70℃

SD-3SC

※5

d

Ⅱ

2G

Ex db IIC T4

Gb-40℃ ≤ Ta ≤ +70℃

SD-3RI
SD-3DRI
GD-3RI
SD-3GH
SD-3DGH
GD-3GH
SD-3GHS
SD-3DGHS
GD-3GHS
SD-3NC
SD-3DNC
GD-3NC
SD-3SP
SD-3DSP
GD-3SP
SD-3EC
SD-3DEC
GD-3EC
SD-3ECS
SD-3DECS
GD-3ECS
SD-3ECB
SD-3DECB
GD-3ECB※1 环境温度是指能够维持防爆性能的温度范围，并非满足产品性能的温度范围。关于使用温度范围，请参阅“10-3 各
种原理的检测对象气体”。
※2 与 GD-3GHS、GD-3SP、GD-3EC、GD-3ECS 组合时※3 与 GD-3GH、GD-3NC 组合时※4 与 GD-3RI 组合时※5 与 GD-3ECB 组合时

17 / 1712. 安全上的重要事项

2-4 安全信息＜电气数据＞
选配件（选择 1 个）
型号

供给电压

功耗

信号输出

无警报继电器无数字通信

SD-3RIDC 24V

SD-3DRI(DC 18V -GD-3RIDC 30VmAx)

SD-3GHDC 24V

SD-3DGH(DC 18V -GD-3GHDC 30VmAx)

SD-3GHSDC 24V

SD-3DGHS(DC 18V -GD-3GHSDC 30VmAx)

SD-3NCDC 24V

SD-3DNC(DC 18V -GD-3NCDC 30VmAx)

SD-3SP

DC 24V

SD-3DSP(DC 18V -GD-3SP

DC 30VmAx)

SD-3ECDC 24V

SD-3DEC(DC 18V -GD-3ECDC 30VmAx)

SD-3ECS

DC 24V

SD-3DECS(DC 18V -GD-3ECS

DC 30VmAx)

SD-3ECBDC 24V

SD-3DECB(DC 18V -GD-3ECBDC 30VmAx)

SD-3SC(DC 18V -3.8WmAx

0 - 22mA + HART―

1.2WmAx――

4.5WmAx

0 - 22mA + HART

―

2WmAx

―

―

4.5WmAx

0 - 22mA + HART

―

2WmAx

―

―

4.5WmAx

0 - 22mA + HART

―

2WmAx

―

―

3.5WmAx

0 - 22mA + HART

―

1WmAx

―

―

2.8WmAx

0 - 22mA + HART

―

1WmAx

―

―

2.8WmAx

0 - 22mA + HART

―

1WmAx

―

―

3.1WmAx

0 - 22mA + HART

―

1WmAx

―

―

5WmAx

0 - 22mA + HART―

DC 24VDC 30VmAx)

警报继电器（接点输出）

数字通信仅电阻负载
DC 30V

1A

AC 250V

2A―

RS485 输出―DC 30V

1A

AC 250V

2A―

RS485 输出―DC 30V

1A

AC 250V

2A―

RS485 输出―DC 30V

1A

AC 250V

2A―

RS485 输出―DC 30V

1A

AC 250V

2A―

RS485 输出―DC 30V

1A

AC 250V

2A―

RS485 输出―DC 30V

1A

AC 250V

2A―

RS485 输出―DC 30V

1A

AC 250V

2A―

RS485 输出―DC 30V

1A

AC 250V

2A

RS485 输出＜证书编号＞
 IECEx PRE 20.0064 X
 Presafe 20 ATEX 69725 X
 DNV 22 UKEX 25924X※ 证书编号末尾的“X”表示以下特殊使用条件。
·请勿进行防爆接合部的修理。
·关于 ATEX 规格，不支持基于 ATEX 指令的附录 II 1.5.5 的测量功能。该指令提供与气体检测仪和安全装置
的性能有关的导则，并符合相关的欧盟统一标准的要求。
・根据 UKSI 2019 年第 1107 号法规附表 1 第 24 段规定的测量功能不包括在本英国类型证书中。 对于气体性能和安全装置的要求，必须满足单独的英国指定标准的要求。
·对窗板施加强烈冲击，可能因破损等导致防爆性能受损。
·本仪器适用机械损伤可能性“低(2J)”的防爆条件。
·在使用、维护和清洁作业时，请勿使用干布擦拭铭牌。可能会累积最高 6nF 的电容。
·在 SD-3 上安装避雷器时，安装螺钉部位使用高强度的螺钉锁固剂。
·关于防爆的使用温度范围，请参阅“2-4 安全信息”中的<防爆性能数据>。
·关于 IECEx 规格，不支持基于 EUT 的测量功能。该指令是与气体检测仪和安全装置的性能有关的导则。
·客户不可更换电池。
18 / 1712. 安全上的重要事项

2-4 安全信息＜适用标准＞
 IEC 60079-0：2017
 IEC 60079-1：2014
 IEC 60079-11：2011
 EN IEC 60079-0：2018
 EN 60079-1：2014
 EN 60079-11：2012
 BS EN IEC 60079-0：2018
 BS EN 60079-1：2014
 BS EN 60079-11：2012＜使用说明书编号＞
 PT2-287

邮编：174-8744 东京都板桥区小豆泽 2-7-6电话：(03)3966-1113
FAX：(03)3558-9110
E-mail：intdept@rikenkeiki.co.jp
主页：https://www.rikenkeiki.co.jp

19 / 1712. 安全上的重要事项

2-4 安全信息＜系统构成图＞

安装图
危险场所(Zone 1)

非危险场所气体检测仪(SD-3)
显示器
POW(+)
COM(-)
Sig(+)
Sig(-)

信号线缆

POW(+)
COM(-)
Sig(+)
Sig(-)

安装图
危险场所(Zone 1)

非危险场所气体检测仪(SD-3)
显示器
POW(+)
COM(-)
Sig(+)
Sig(-)

RLY

信号线缆

信号线缆

20 / 171

POW(+)
COM(-)
Sig(+)
Sig(-)RLY

2. 安全上的重要事项

2-4 安全信息

安装图
危险场所(Zone 1)

非危险场所气体检测仪(SD-3(RS485))
显示器
POW(+)
COM(-)
Sig(+)
Sig(-)

信号线缆

RS485

信号线缆

POW(+)
COM(-)
Sig(+)
Sig(-)

RS485

安装图
危险场所
(Zone 1)

非危险场所气体检测仪(SD-3SC)显示器

遥控传感器头
(GD-3)
POW(+)
COM(-)
RS485(Y)RS485(Z)信号线缆

POW(+)
COM(-)
Sig(+)
Sig(-)

信号线缆

POW(+)
COM(-)
RS485(Y)RS485(Z)21 / 171

POW(+)
COM(-)
Sig(+)
Sig(-)

2. 安全上的重要事项

2-4 安全信息

安装图
危险场所(Zone 1)

非危险场所气体检测仪(SD-3SC)显示器

遥控传感器头
(GD-3)
POW(+)
COM(-)
RS485(Y)RS485(Z)信号线缆

POW(+)
COM(-)
Sig(+)
Sig(-)

信号线缆

POW(+)
RLY
COM(-)
RS485(Y)RS485(Z)信号线缆

POW(+)
COM(-)
Sig(+)
Sig(-)

RLY

安装图
非危险场所

危险场所
(Zone 1)气体检测仪(SD-3SC)

显示器
POW(+)
COM(-)
Sig(+)
Sig(-)

信号线缆

遥控传感器头
(GD-3)
POW(+)
COM(-)
RS485(Y)RS485(Z)

信号
线缆

RS485
POW(+)
COM(-)
RS485(Y)RS485(Z)POW(+)
COM(-)
Sig(+)
Sig(-)

RS485

信号线缆

22 / 1712. 安全上的重要事项

2-4 安全信息

注意• 安装时请遵从以下安装要求。
• 请勿拆解、改造、变更本仪器。
• 进行重新调试（包括气体校正）或更换部件时，请联系经销商或就近的本公司营业网点。
• 请使用线缆衔套进行接地线或接合导体的连接。此外，请安装接合导体，使其不会松动或扭曲。
• 大气中存在爆炸性气体时，请勿打开盖子。
• 流路及传感器部内的最大压力为 10.13kPa（表压）。
• 为本仪器接通电源前，请务必连接到接地端子。
• 接地处理请按照 D 种接地标准（接地电阻低于 100Ω）。
• 请使用压接端子，将接地线连接到单元内部的接地端子。此外，外部接地端子请使用截面积 4mm2 以上的线缆。
• 安装时请使用获认证的线缆防水接头。
• 将本仪器安装在环境温度超过 65℃的场所时，请选择并使用可在比最高环境温度还高至少 5℃的温度下使用的线缆。
• 请勿向盖子的窗口施加强压力或冲击。否则可能因损坏而丧失防爆性能。
（已通过测试确认机械危险性很低（2 焦耳）。
）• 使用转换适配器时，请勿对 1 个连接口堆叠使用 2 个或更多转换适配器。

注意
为抽吸式时
• 使用时请控制流量为 0.4 以上~1.5L/min 以下。

23 / 1712. 安全上的重要事项

2-4 安全信息＜连接口的螺钉种类和尺寸(SD-3、SD-3SC)＞
连接口 3

连接口 4

连接口 2

连接口 1

连接口 5

连接口 6
＜标准＞
连接口 1

连接口 2

连接口 3

连接口 4

连接口 5

连接口 6

M25 × 1.5

M25 × 1.5―――

M25 × 1.5

 “―”表示没有连接口。
 连接口 1、2、6 可用作线缆插入口。但是，以线缆连接主机(SD-3SC)和遥控传感器头(GD-3)时，请使用连接口 6。连接遥控传感器头(GD-3)时，请使用获认证的线缆防水接头。
 未使用的连接口请使用堵头塞住。
 各连接口的详细用途请参阅“4-7 线缆的连接”。
＜选配件＞备有对连接口 4、5 进行了螺纹加工的以下选配件系列。
连接口 1

连接口 2

连接口 3

连接口 4

连接口 5

连接口 6

M25 × 1.5

M25 × 1.5―

M25 × 1.5

M25 × 1.5

M25 × 1.5

 “―”表示没有连接口。
 连接口 1、2、4、6 可用作线缆插入口。但是，以线缆连接主机(SD-3SC)和遥控传感器头(GD-3)时，请使用连接口 6。连接遥控传感器头(GD-3)时，请使用获认证的线缆防水接头。
 可同时使用 2 个外部线缆连接口。
（连接遥控传感器头(GD-3)的线缆除外）
 连接口 2 或连接口 4 可以使用选配的转换适配器（将 M25 × 1.5 螺纹转换成 1/2NPT 螺纹的适配器）连接选配的避雷器。
 连接口 5 使用选配的 HART 调制解调器连接器进行 HART 通信。请勿用于插入外部线缆或连接避雷器。不使用 HART 通信时，请使用堵头塞住。
 未使用的连接口请使用堵头塞住。
 各连接口的详细用途请参阅“4-7 线缆的连接”。

24 / 1712. 安全上的重要事项

2-4 安全信息＜可连接的组件(SD-3、SD-3SC)＞
可与主机(SD-3、SD-3SC)连接的组件如下所示。
连接口 1

连接口 2

连接口 3

连接口 4

连接口 5

连接口 6

传感器总成

×

×

×

×

×

〇

EC 安全栅

×

×

×

×

×

〇

认证线缆防水接头

〇

〇

×

〇

×

〇

M25 插头

〇

〇

×

〇

〇

×

转换适配器
(M25⇔1/2NPT)

〇

〇

×

〇

×

〇

转换适配器
(M25⇔3/4NPT)

〇

〇

×

〇

×

〇

转换适配器
(M25⇔M20)

〇

〇

×

〇

×

〇

避雷器

×

〇

×

〇

×

×

〇：可连接
×：不可连接注记 请连接适合主机(SD-3、SD-3SC)各连接口的组件。
 请按照 30N·m 及以上的扭矩值紧固 M25 插头和转换适配器。
 请确保螺钉啮合的螺纹牙数至少有 5 牙。

＜关于避雷器＞
制造商
部件
名称/编号
防爆性能
编号及版本
认证
编号

额定
安装

适用标准
电压/电流
温度※
安装螺钉

Eaton Electric Limited
浪涌保护器 TP48 及 TP32 系列
II 2G Ex db IIC T6 Gb
IECEx BAS 15.0056X Issue 1 Baseefa 04 ATEX 0053X
EN 60079-0:2012+A11:2013
EN 60079-1:2014
IEC 60079-0:2011 Edition:6.0
IEC 60079-1:2014 Edition:7.0
DC 48V-40 - +70℃
1/2NPT※ 使用浪涌保护器时，SD-3/GD-3 的环境温度下限为-40℃。

25 / 1712. 安全上的重要事项

2-4 安全信息

证书编号末尾的“X”表示以下特殊使用条件。
认证编号

IECEx BAS 15.0056X
Baseefa 04 ATEX 0053X“X”固有的使用条件

判定为妥善保护产品，避免因拉扯连接线缆等导致机械损伤，请用符合使用条件的端子或在接合部内进行尾端处理。

总成符合这些装置不配备接地或接合导体的外部连接器。请用户自行负责安装，确保充分的接地导通。

总成符合该装置也提供“Ex i”版本，带有两者的标志。请用户自行
判断防爆性能。

不适用

安装到耐压防爆外壳时，请为浪涌保护器的安装螺钉涂抹高强度的锁固剂。

“X”标志

26 / 1712. 安全上的重要事项

2-4 安全信息＜可连接的组件(GD-3)＞

连接口 1

连接口 2

连接口 3

可与遥控传感器头(GD-3)连接的组件如下所示。
连接口 1

连接口 2

连接口 3

传感器总成

×

×

〇

EC 安全栅

××〇认证线缆防水接头〇〇×M25 插头

〇

〇

〇

转换适配器
(M25⇔1/2NPT)

〇

〇

〇

转换适配器
(M25⇔3/4NPT)

〇

〇

〇

转换适配器
(M25⇔M20)

〇

〇

〇

〇：可连接
×：不可连接注记 请连接适合遥控传感器头(GD-3)各连接口的组件。
 请按照 30N·m 及以上的扭矩值紧固 M25 插头和转换适配器。
 请确保螺钉啮合的螺纹牙数至少有 5 牙。

27 / 1712. 安全上的重要事项

2-4 安全信息＜连接示例＞

SD-3SCGD-3RI

SD-3SCGD-3ECB

28 / 1712. 安全上的重要事项

2-4 安全信息＜拆解图＞＜SD-3＞

容器

盖子
堵头

堵头

气体传感器

传感器护板

＜GD-3＞

容器

盖子
堵头

气体传感器

传感器护板

29 / 1712. 安全上的重要事项

2-4 安全信息＜容器背面的螺孔(SD-3)＞
主机容器的背面有 4 个螺孔。
 螺钉尺寸：M5 螺距：0.8
 螺钉深度：5 公螺纹的最大允许长度：LmAx = 4.2mm + x（“x”为中间件的厚度）

容器背面的螺孔（4 处）＜容器背面和底面的螺孔(GD-3)＞
主机容器的背面有 4 个螺孔，底面也有 4 个螺孔。
 螺钉尺寸：M5 螺距：0.8
 螺钉深度：5 公螺纹的最大允许长度：LmAx = 4.2mm + x（“x”为中间件的厚度）＜容器背面＞＜容器底面＞

容器背面的螺孔（4 处）

容器底面的螺孔（4 处）

30 / 1713. 产品组成

3-1 主机及附件

3产品组成
3-1 主机及附件
请打开包装箱，确认主机及附件。
如有缺失，请咨询经销商或就近的本公司营业网点。

3-1-1 主机(SD-3)＜扩散式＞＜无 EC 安全栅规格＞＜有 EC 安全栅规格＞＜无 EC 安全栅规格＞＜有 EC 安全栅规格＞＜抽吸式＞

31 / 1713. 产品组成

3-1 主机及附件

3-1-2 主机(SD-3SC)＜主机＞

3-1-3 遥控传感器头(GD-3)＜遥控传感器头＞＜无 EC 安全栅规格＞＜有 EC 安全栅规格＞

32 / 1713. 产品组成

3-1 主机及附件

3-1-4 标准附件
标准附件

数量

部件编号

说明

控制钥匙

因交货
台数
而异

4286 9200 80

操作本仪器时使用的控制
钥匙。

操作杆

1个

2594 0481 90

将线缆连接到端子排时使用的
工具。

六角棒扳手
（对边 2）

因交货台数
而异1510 5020 40

紧固内六角固定螺钉(M4)时
使用的工具。

使用说明书

1本

注记
 控制钥匙和六角棒扳手的随附数量因交货台数而异。
1 - 10 台：1 个
11 - 20 台：2 个
21 - 50 台：3 个
51 台及以上 ：4 个

注意
• 操作本仪器时，请使用随附的控制钥匙。如果使用非随附的控制钥匙，可能无法正常操作。
• 随附的控制钥匙以非常强力的磁铁制成。如果靠近信用卡或 IC 卡等磁性产品，存储数据可能会受损。

33 / 1713. 产品组成

3-1 主机及附件

3-1-5 特别附件（选配件）特别附件（选配件）

部件编号

校正盖
（IRF 传感器用）

校正盖
（NCF/SGF/SHF 传感
器用）校正盖（ESF 传感器用）

4283 9011 00

SD-3RI 用的校正盖。
＜材质＞
主体：PP
连接嘴：不锈钢/特氟龙4283 9012 70

SD-3NC/GH/GHS/SP 用的
校正盖。
＜材质＞
主体：PP
连接嘴：不锈钢/特氟龙4283 9013 40

SD-3EC/ECS/ECB 用的
校正盖。
＜材质＞
主体：PP
连接嘴：不锈钢/特氟龙墙面用4283 9019 80
保护罩
管用4283 9045 60

防晒罩

防溅罩
（IRF 传感器用）

34 / 171说明保护 SD-3 主机的罩盖。
＜材质＞
SUS3164283 9015 90

避免阳光直射或辐射热导致
温度上升的罩盖。
＜材质＞
SUS3164283 9016 60

用于 SD-3RI，保护传感器免受
水溅、尘染的罩盖。
底面刻有识别编号“1”。
＜材质＞
树脂制

3. 产品组成

3-1 主机及附件

特别附件（选配件）

部件编号

说明

4283 9017 30

用于 SD-3NC/GH/GHS/SP，
保护传感器免受水溅、尘染的
罩盖。
底面刻有识别编号“2”。
＜材质＞
树脂制

防溅罩
（ESF 传感器用）

4283 9031 80

用于 SD-3EC/ECS/ECB，
保护传感器免受水溅、尘染的
罩盖。
底面刻有识别编号“3”。
＜材质＞
树脂制

堵头

4283 9018 10

M25 × 1.5 用的堵头。
＜材质＞
等同 SUS316

4283 9058 70

将 M25 × 1.5 螺纹转换成
1/2NPT 螺纹的适配器。
＜材质＞
等同 SUS316

4283 9057 00

将 M25 × 1.5 螺纹转换成3/4NPT 螺纹的适配器。
＜材质＞
等同 SUS316

4283 9047 10

将 M25 × 1.5 螺纹转换成
M20 × 1.5 螺纹的适配器。
＜材质＞
等同 SUS316

4283 0030 20

用于 SD-3NC/GH/GHS/SP 的过滤器，可去除大气中微量的硅。配备后可以延长传感器的使用寿命。※2

防溅罩（NCF/SGF/SHF 传感
器用）

转换适配器

转换适配器

转换适配器

除硅过滤器
（型号：SI-8）※1（NCF/SGF/SHF 传感
器用）35 / 1713. 产品组成

3-1 主机及附件

特别附件（选配件）

部件编号

说明

4283 0040 10

用于 SD-3NC/GH/GHS/SP 的过滤器，可去除大气中微量的硅。配备后可以延长传感器的使用寿命。
其除硅能力优于除硅过滤器。※3
此外，也可以用于去除干扰气
体。配备后可以抑制检测对象
气体以外的干扰影响。

线槽安装套件4264 9580 80

用于将遥控传感器头(GD-3)
安装到线槽的套件。
＜材质＞
SUS316

HART 通信用
线缆2905 2439 10

连接 HART 调制解调器时使用
的中继线缆。

避雷器（3 线连接用）(TP48-3-N-NDI)

4283 9055 50

避雷器用于限制因雷击引起的瞬时过电压。带转换适配器(M25×1.5→1/2NPT)。

避雷器（4 线连接用）(TP48-4-N-NDI)

4283 9056 20

避雷器用于限制因雷击引起的瞬时过电压。带转换适配器(M25×1.5→1/2NPT)。

保险丝

2564 0125 10

Littelfuse 保险丝。
(1.25A DC 100V)

4283 9046 30

将主机安装到垂直杆（尺寸：50A(2B)）时使用的 U 型螺
栓、弹簧垫圈和六角螺母。
＜材质＞
SUS304活性碳过滤器（型号：CF-8304）※1（NCF/SGF/SHF 传感
器用）

U 型螺栓部件※1 使用过滤器时，建议配备防溅罩。
※2 配备除硅过滤器时，检测对象气体受到限制。
※3 与配备除硅过滤器时相比，检测对象气体受到限制。

36 / 1713. 产品组成

3-2 各部名称与作用

3-2 各部名称与作用＜SD-3＞＜显示部（SD-3、SD-3SC 通用）＞⑥ ⑦ ⑧ ⑨ ⑩

①

④
⑭

②
＜SD-3SC＞
③

⑤

⑪ ⑫

编号
①

名称
MENU/ESC 键②③

7 段数码管 LED 显示
SET 键

④
⑤

▲键
▼键

⑥

电源指示灯
(POWER)

⑦
⑧
⑨
⑩
⑪
⑫
⑬
⑭

警报灯(ALM1)
红外端口
警报灯(ALM2)
故障灯(FAULT)PPM 指示灯%LEL 指示灯
%指示灯
传感器部

⑬

功能
在检测模式下长按此键，即切换到用户模式。
在用户模式、维护模式下，用于取消设定或返回上 1 个画面。
在用户模式、维护模式长按，就返回检测模式。
显示气体浓度等。
在检测模式下按此键，显示 ALM1（第一警报）、ALM2（第二警报）的警报设定值。
在用户模式、维护模式下，用于确定值或选择设定。
用于在用户模式、维护模式下切换设定项目和数值的调整(UP)。
在用户模式下用于菜单的切换、数值的调整(DOWN)。
在检测模式下按此键，显示外部输出。
电源灯。接通电源时点亮（绿色）。
在用户模式、维护模式下闪烁（绿色）。
ALM1（第一警报）的警报灯。达到 ALM1（第一警报）的警报设定值就会
点亮（红色）。
IrDA 通信端口。
ALM2（第二警报）的警报灯。达到 ALM2（第二警报）的警报设定值就会
点亮（红色）。
这是故障灯。本仪器发生异常就会点亮（黄色）。
显示 ppm 浓度范围内的气体浓度值时点亮（绿色）。
显示%LEL 浓度范围内的气体浓度值时点亮（绿色）。
显示 vol%浓度范围内的气体浓度值时点亮（绿色）。
搭载了检测气体的传感器。

37 / 1713. 产品组成

3-3 框图

3-3 框图
本仪器的框图（电气系统图）如下所示。

＜SD-3＞

电源供给部DC 24V显示部(LED)·POWER·ALARM1·ALARM2·FAULT
·ppm
·%LEL
·vol%

电源电压转换部

外部输出部
4 - 20mA with HART

HART 调制解调器部

警报接点输出部（选配件）1.气体警报输出(ALARM1)
2.气体警报输出(ALARM2)
3.故障输出(FAULT)
接点动作
常规测量时
警报时

控制部
CPU

显示部
7 段数码管 LED

磁性操作部·MENU/ESC
·SET
·▲
·▼

非励磁
励磁

传感器

38 / 1713. 产品组成

3-3 框图

＜SD-3SC＞
电源供给部
DC 24V显示部(LED)·POWER·ALARM1·ALARM2·FAULT
·ppm
·%LEL
·vol%

电源电压转换部

外部输出部
4 - 20mA with HART

HART 调制解调器部

警报接点输出部（选配件）1.气体警报输出(ALARM1)
2.气体警报输出(ALARM2)
3.故障输出(FAULT)

控制部
CPU

显示部
7 段数码管 LED

磁性操作部·MENU/ESC
·SET
·▲
·▼

接点动作
常规测量时 非励磁警报时 励磁

GD-3＜GD-3＞

SD-3SC
控制部
CPU电源电压转换部

39 / 171传感器

4. 安装方法

4-1 有关安装的注意事项

4

安装方法
4-1 有关安装的注意事项选择适当的安装场所对系统性能而言非常重要。
本仪器的安装作业务必遵守以下注意事项。
如果不遵守这些注意事项，仪器可能会发生故障，无法正常工作。

注意• 本仪器是精密仪器。由于安装场所（环境）不同，本仪器可能无法发挥性能。请确认安装场所的环境，视情况采取必要的措施。
• 本仪器在安全防灾上起着重要的作用，因此请在有效的场所安装所需的数量。
• 根据气体种类和作业区域，气体易泄漏或易积聚的位置会有所不同。请充分考虑场所和数量进行安装。
• 杂质、雪、泥、外部设备等不得阻挡气体或蒸气与传感器的接触。保护等级并不能保证本仪器在暴露于这
些条件的期间内及之后能够正常检测到气体。灰尘沉积、浸没或溅到高压的水后，本仪器应实施动作确认
和校正。
• 关于检测仪的安装方向，建议让传感器朝向下方。如果能保护传感器免受机械损伤、污渍、浸水，以其他
朝向安装也没有问题。
• 特别附件请使用正品。

4-1-1 有关安装的一般信息
在安装作业中请注意以下几点。
 将本仪器安装在环境温度超过 65℃的场所时，请选择并使用可在比使用时设想的最高环境温度还高至少 5℃的温度下使用的线缆。
 SD-3、SD-3SC 的主机有 2 个 M25 × 1.5 的连接口，用于现场配线。不使用连接口时，请务必使用堵头塞住。
 GD-3 有 2 个用于连接 SD-3SC 的 M25×1.5 的连接口。不使用的任意一边连接口，请务必使用堵头塞住。
 请勿安装工作原理与本型号不同的传感器。
 使用绞合导体时，请在前端部使用环箍。
 请通过绝缘电源供给二次电路。
（不适用于警报继电器接点。
）
 不使用导管进行安装时，请使用符合防爆检验的线缆防水接头。此外，请将线缆的屏蔽连接到主机、线缆防水
接头、指示器等处，以增强对电磁波噪声等外部影响的耐受力。
 连接警报继电器的配线（选配件），请根据额定电压、电流和环境条件选择连接线缆、端子排等零件。
此外，安装时请充分考虑以下内容。
 对安装气体检测仪系统产生的所有影响 与安装气体检测仪系统有关的规定和规则 与气体检测仪系统的工作、电源和信号线连接有关的规则 周围环境对气体检测仪系统的影响 与检测气体的物理特性、环境空气的运动和流动、潜在泄漏等有关的详细情况
40 / 1714. 安装方法

4-1 有关安装的注意事项 所需维护的便利性和所用附件的形状

4-1-2 有关安装的限制信息＜不应安装在有振动冲击的场所＞
本仪器由精密的电子部件构成。请安装在没有振动、冲击等，没有坠落等风险的、稳定的场所。

＜不应安装在水、油、化学药剂等可能溅落的场所＞
安装请避开水、油、化学药剂等液体可能溅落的场所。如需安装在可能受此影响的场所，请使用选配的保护罩（保护主机）或防溅罩（保护传感器）。

＜不应安装在超过使用温度范围的场所＞关于本仪器的使用温度范围，请参阅“10-3 各种原理的检测对象气体”。
请安装在使用时不会超过使用温度范围的稳定的场所。

＜不应安装在日光直射的场所或温度剧变的场所＞
请避开日光直射或有辐射热（高温物体放射的红外线）的场所、可能导致仪器温度剧变的场所。否则可能导致温度
上升超过指定阈值、仪器内部结露，或者无法适应剧烈的温度变化。
如果在日光直射的场所使用，建议使用选配的防晒罩。

＜不应安装在无法维护的场所、作业有危险的场所＞
本仪器需要定期进行维护。
请勿安装在无法维护的场所或作业可能发生危险的场所。
 装置内部等，维护时需停止装置工作的场所 不拆下装置的一部分就无法进行维护的场所 因配管、线槽等的存在而无法拆下本仪器的场所 高压线附近等，实施维护作业时有危险的场所＜不应安装在接地施工不完备的装置外壳上＞
如果安装在装置外壳上，请切实做好与 D 种等同的接地施工。

＜不应安装在周围有杂质气体的场所＞
请勿安装在周围有以下气体的场所。
传感器的类型

降低传感器气体敏感度的气体

腐蚀传感器的气体·有机硅气体（D4 硅氧烷、D5 硅氧烷等）·硫气（SOX、H2S 等）·聚合物质（环氧乙烷、丙烯腈、丁二烯、苯乙烯等）·催化剂毒物（磷化合物、卤代烃、金属蒸气等）·影响程度因传感器型号而异。

·腐蚀性气体（SOX、NOX等）·酸性气体（HF、HCl 等）

SGF

SHF

NCF
IRF

无

ESF

无

无

41 / 1714. 安装方法

4-1 有关安装的注意事项

＜安装环境存在干扰气体时的注意事项＞
安装环境存在如下干扰气体时请注意。
传感器的类型干扰气体SGF

对象气体以外的碳化氢(HydroCarbon)、醇类、有机溶剂、H2O 等

SHF
NCF

对象气体以外的碳化氢(HydroCarbon)、醇类、有机溶剂等IRF

可燃性传感器：对象气体以外的碳化氢(HydroCarbon)、醇类、有机溶剂等ESF因传感器而异。（例如 CO 传感器为 H2 等）＜应使用适当的电源和电线＞
本仪器的运行需要 DC 24V(DC 18V - DC 30V)的电源。供电的电线请使用 1.308mm2 (AWG16)或 2.082mm2(AWG14)。
请使用带屏蔽的 3 线线缆。

＜应远离噪声源仪器（主机及线缆）＞请避开周围有高频仪器、高电压仪器的场所进行安装。否则可能会受到电磁波噪声的影响。

＜4 - 20mA 信号输出的阻抗应设定为适当的值＞
为保证正常运行，4 - 20mA 信号输出的阻抗需在 600Ω以下使用。这也取决于对本仪器的供给电压。此外，请根据本仪器的运行电压或阻抗(HART)，遵守特定的最小或最大阻抗。
＜因供给电压差异引发的 4 - 20mA 信号输出的阻抗降额＞

4 - 20mA出力負荷ディレーティング
4 - 20mA 输出负载降额
700 Ω
600 Ω
500 Ω
400 Ω
300 Ω
200 Ω
100 Ω
0Ω
30 V 29 V 28 V 27 V 26 V 25 V 24 V 23 V 22 V 21 V 20 V 19 V 18 V 17 V 16 V 15 V

＜使用 HART 通信时的 4 - 20mA 信号输出的阻抗范围＞
配备 HART 接口的仪器可以在 HART 通信或 HART 多点模式下操作。
运行模式基于电源电压的信号的阻抗范围

无 HART 通信的操作

DC 18V，最大 300Ω
DC 23V，最大 600Ω或直线上升

基于 HART 通信的操作（HART 模式）

DC 18V - DC 30V 的范围内，最低 250Ω

HART 多点操作

DC 18V，最大 300Ω
DC 23V，最大 600Ω或直线上升
DC 18V - DC 30V，250 - 600Ω

42 / 1714. 安装方法

4-1 有关安装的注意事项＜正确使用警报接点＞
本仪器的警报接点的用途是作为运行外部蜂鸣器或警报指示灯、旋转灯等的信号传递手段。请勿用于涉及人命的
控制用途（例如断路阀的控制）。
此外，请定期确认开闭运行，包括进行定期检查。如果长期没有开闭运行，在某些使用环境下可能无法正常运行。
本仪器的警报接点规格是基于电阻负载条件的规格。在警报接点上使用感应负载时，接点部会产生逆电动势，
因此容易发生以下故障。
 警报继电器接点部的熔敷、绝缘不良、接触不良 在本仪器的内部产生高电压导致不特定电气部件损坏 CPU 失灵导致异常运行运行负载时，为稳定本仪器的运行，保护警报接点，请采取以下正确措施。
 请用外部继电器（低电压 AC 100V 以内）进行中继（接点放大）。此时，请在外部继电器上也安装符合额定值的电涌吸收部件 SK1。
 在外部继电器的负载侧，也请视需要安装电涌吸收部件 SK2。
 根据负载条件，电涌吸收部件有时也可以装在接点侧。请确认负载的运行，安装在适当的场所。
＜SD-3＞

负载

线圈
电源

电源
警报触点
外部继电器
（低电压继电器）※危险• 请绝对不得用于涉及人命的控制用途（例如断路阀的控制）。

43 / 171

SK1、SK2：电涌吸收部件

4. 安装方法

4-1 有关安装的注意事项警告• 如果长期没有开闭运行，可能无法正常运行。请定期确认接点的开闭运行。
• 如果长期暴露在有机气体或硫化气体环境下，可能引发接触不良等问题。
• 原则上请勿用本仪器的警报接点运行感应负载。特别是请绝对不要用于荧光灯、马达等的运行。
• 运行感应负载时，请用外部继电器进行中继（接点放大）。但是，外部继电器的线圈也属于感应负载，因此请使用低电压（AC 100V 以内）驱动的继电器，用正确的电涌吸收部件（CR 电路等）保护本仪器的接点。
• 请勿在硅环境下使用。

注意• 无励磁状态的 b 接点（断开接点）由于外力等物理冲击，可能发生瞬时打开（开路）运行。警报接点使用b 接点时，请考虑可能发生瞬时运行，采取在 b 接点接收侧增加信号延迟运行（1 秒左右）等措施。

44 / 1714. 安装方法

4-2 系统设计上的注意事项

4-2 系统设计上的注意事项
本仪器的系统设计注意以下几点。

注意• 不稳定的电源、噪声会引起误动作、误警报。
• 设计使用本仪器的系统时，请反映出本节所述的内容。

4-2-1 使用稳定的电源
在接通电源时或瞬时停电时，在系统稳定之前，本仪器的外部输出和警报接点可能会启动。因电源问题等导致外部输出或警报接点启动时，请采取适当的措施，例如使用安全电源。
请为本仪器提供以下电源。
电源电压
DC 24V(DC 18～DC 30V)：主机端子电压瞬时停电容许时间
最多约 20ms※1
处理示例：为了保证连续运行及正常运行，请在外部
安装无停电电源装置等。
其他请勿与包含大电力负载或高频噪声处理示例：
的电源共用。
请根据需要使用线路滤波器等仪器，与噪声源隔离。
※1 瞬时停电至少 20ms 后复位时，请重新启动。

4-2-2 防雷措施
在有如下配线的工厂和成套设备中，连接到仪器的线缆变成雷电的接收天线，可能会被雷击。
 在室外配线时 与从室外接入的线缆在同一线槽内平行配线时一旦发生雷击，连接到线缆的仪器可能会被雷击的能量破坏。此外，即使将线缆放入金属管中或者埋入地下，也无
法完全防止雷击引发的感应雷电涌。虽然无法完全杜绝雷击灾害，但是也有如下措施。

＜防雷措施＞
请根据设备的重要程度及环境采取妥善措施。
包括在现场仪器及中央处理器前面安装避雷器（线缆安全器），作为防止万一感应雷电涌落到线缆上的措施。使用方法请咨询避雷器制造商。

＜接地处理＞
雷电及其他现象也会引发浪涌噪声。为保护仪器免受这些因素的影响，请将仪器接地。

注记 避雷器内置了消除浪涌电压的电路，用以防止现场仪器损坏。因此安装避雷器可能会使信号衰减。
安装避雷器时，请事先确认运行后使用。此外，避雷器不能消除所有的浪涌电压。

45 / 1714. 安装方法

4-3 确保维护空间

4-3 确保维护空间
安装本仪器时，请确保维护时打开/关闭盖子所需的空间。

＜主机(SD-3、SD-3SC)＞
1200

300

※ 图为 SD-3。SD-3SC 的维护空间也相同。

＜遥控传感器头(GD-3)＞
1100

300

46 / 1714. 安装方法

4-3 确保维护空间

注意
本仪器需要定期进行维护。
请勿安装在无法维护的场所或作业可能发生危险的场所。
• 装置内部等，维护时需停止装置工作的场所• 不拆下装置的一部分就无法进行维护的场所• 因配管、线槽等的存在而无法拆下本仪器的场所• 高压线附近等，实施维护作业时有危险的场所47 / 1714. 安装方法

4-4 安装主机

4-4 安装主机
4-4-1 安装到垂直杆使用 2 个 U 型螺栓，将主机(SD-3、SD-3SC)安装到垂直杆（尺寸：50A(2B)）
。
1 将 U 型螺栓插入垂直杆，穿过主机上侧的槽口
2 用 2 个垫圈、2 个弹簧垫圈、2 个螺母(M10)固定 U 型螺栓和主机
3 将第 2 个 U 型螺栓插入垂直杆，穿过主机下侧的槽口
4 用 2 个垫圈、2 个弹簧垫圈、2 个螺母(M10)固定 U 型螺栓和主机

垂直杆

U 型螺栓

上侧的槽口

下侧的槽口

※ 图为 SD-3。SD-3SC 的安装方法也相同。

48 / 1714. 安装方法

4-4 安装主机

4-4-2 安装到墙面
将主机(SD-3、SD-3SC)安装在墙面上。
1 将主机推靠在墙面上，用 4 个双垫圈螺钉(M10)固定
螺钉位置

螺钉位置
※ 图为 SD-3。SD-3SC 的安装方法也相同。

注意
本仪器需要定期进行维护。
请勿安装在无法维护的场所或作业可能发生危险的场所。
• 装置内部等，维护时需停止装置工作的场所• 不拆下装置的一部分就无法进行维护的场所• 因配管、线槽等的存在而无法拆下本仪器的场所• 高压线附近等，实施维护作业时有危险的场所49 / 1714. 安装方法

4-4 安装主机

4-4-3 安装保护罩
将保护罩安装到主机(SD-3、SD-3SC)上。

＜安装到垂直杆（尺寸：50A(2B)）＞
1

将底座和角铁安装到主机，如右图所示，从背面用 4 个双垫圈螺钉(M5 × 6)固定

底座

角铁

2

将 U 型螺栓插入垂直杆，穿过主机上侧的槽口，
用 2 个垫圈、2 个弹簧垫圈、2 个螺母(M10)固定

3

将 U 型螺栓插入垂直杆，穿过主机下侧的槽口，
用 2 个垫圈、2 个弹簧垫圈、2 个螺母(M10)固定

U 型螺栓

扣爪

4

将保护罩对准底座顶部的 2 处扣爪进行安装5紧固保护罩左右的滚花螺钉

保护罩

滚花螺钉
※ 图为 SD-3。SD-3SC 的安装方法也相同。

50 / 1714. 安装方法

4-4 安装主机＜安装到墙面上＞
1

用 4 个双垫圈螺钉(M10)共同紧固主机和底座

底座

2

将保护罩对准底座顶部的 2 处扣爪进行安装3紧固保护罩左右的滚花螺钉

扣爪
保护罩

滚花螺钉

※ 图为 SD-3。SD-3SC 的安装方法也相同。

51 / 1714. 安装方法

4-4 安装主机

4-4-4 安装防晒罩
将防晒罩安装到主机(SD-3、SD-3SC)上。

＜安装到垂直杆（尺寸：50A(2B)）＞
1

将 U 型螺栓插入垂直杆，穿过主机上侧的槽口，用底座和 2 个垫圈、2 个弹簧垫圈、2 个螺母(M10)固定2

将 U 型螺栓插入垂直杆，穿过主机下侧的槽口，
用 2 个垫圈、2 个弹簧垫圈、2 个螺母(M10)固定

底座

U 型螺栓

垂直杆

3

将防晒罩对准底座顶部的 2 处扣爪进行安装4紧固防晒罩左右的滚花螺钉

扣爪
防晒罩

滚花螺钉

※ 图为 SD-3。SD-3SC 的安装方法也相同。

52 / 1714. 安装方法

4-4 安装主机＜安装到墙面上＞
1

用 4 个双垫圈螺钉(M10)共同紧固主机和底座

2

将防晒罩对准底座顶部的 2 处扣爪进行安装3

紧固防晒罩左右的滚花螺钉

底座

扣爪
防晒罩

滚花螺钉

※ 图为 SD-3。SD-3SC 的安装方法也相同。

53 / 1714. 安装方法

4-5 安装遥控传感器头(GD-3)

4-5 安装遥控传感器头(GD-3)
4-5-1 安装到墙面
1 将遥控传感器头(GD-3)推靠在墙面上，用 2 个双垫圈螺钉(M6)固定螺钉位置

4-5-2 安装到线槽
将遥控传感器头(GD-3)安装到线槽。
安装到线槽时，使用线槽安装套件。
线槽安装套件尺寸如下。
遥控传感器头(GD-3)

1

切割线槽的顶板开孔
孔的大小请参阅右图。

线槽顶部

100

180

54 / 1714. 安装方法

4-5 安装遥控传感器头(GD-3)

2

将遥控传感器头(GD-3)插入线槽安装套件，用 4 个螺钉(M5)固定
请将采集管连接到遥控传感器头(GD-3)的传感器部，
将其拉出线槽安装套件的顶部，以便能进行气体校正。

3将线槽安装套件插入线槽的孔，用 6 个螺钉(M6)固定线槽安装套件注记 将遥控传感器头(GD-3)安装到线槽内时，请避免传感器部干扰线槽内壁。

线槽55 / 1714. 安装方法

4-6 接地端子的连接

4-6 接地端子的连接
警告• 为本仪器接通电源前，请务必接地。
• 为确保仪器稳定运行和安全，请务必接地。此外，接地线切勿连接到气体管道。
• 请按照 D 种接地标准（接地电阻 100Ω以下）连接接地。
• 接地线上使用线缆衔套安全接地，以免松动或扭曲。
• 请勿与高频仪器或高电压仪器在同一场所或在其周围接地。
请连接起接地螺栓和客户的接地端子。
＜主机(SD-3)＞＜主机(SD-3SC)＞

接地螺栓(M4)

接地螺栓(M4)

注记 接地螺栓带有的标记。

56 / 171＜遥控传感器头(GD-3)＞

接地螺栓(M4)4. 安装方法

4-7 线缆的连接4-7 线缆的连接4-7-1 主机的线缆和仪器的连接口
主机的线缆和仪器的连接口如下所示。
＜SD-3＞＜选配件＞·电源线缆、
信号线缆
mAx 2.0mm2（屏蔽）·避雷器(TP-48)·电源线缆、
信号线缆
mAx 2.0mm2（屏蔽）·避雷器(TP-48)

接触输出
Max 2.0 mm2（屏蔽）＜选配件＞·HART 调制解调器用连接器＜SD-3SC＞·电源线缆、
信号线缆
mAx 2.0mm2（屏蔽）·避雷器(TP-48)＜选配件＞·电源线缆、
信号线缆
mAx 2.0mm2（屏蔽）·避雷器(TP-48)＜选配件＞·HART 调制解调器用连接器（不可在防爆区域内
使用）

接触输出
Max 2.0 mm2（屏蔽）·遥控传感器头(GD-3)的连接线缆Max 2.0mm2（屏蔽双绞线）

57 / 1714. 安装方法

4-7 线缆的连接注记 连接遥控传感器头(GD-3)时，请使用获认证的线缆防水接头。
 可同时使用 2 个外部线缆连接口。
（连接遥控传感器头(GD-3)的线缆除外）请使用获认证的线缆防水接头进行连接。
 未使用的连接口请使用堵头塞住。
 使用转换适配器时，请勿对 1 个连接口堆叠使用 2 个或更多转换适配器。
 转换适配器不应与堵头一起使用。

58 / 1714. 安装方法

4-7 线缆的连接

4-7-2 将线缆连接到主机
将电源线缆、信号线缆和接点用线缆连接到主机(SD-3、SD-3SC)。
1

旋松主机盖子的 1 个内六角固定螺钉(M4)，然后逆时针转动盖子将其取下
请使用六角棒扳手（对边 2）等工具旋松内六角固定螺钉(M4)。
旋松并取下盖子时，请注意避免盖子掉落。

2

旋松 2 个固定显示单元的螺钉(M3)3取出显示单元，挂在壳体的突起部分
显示单元和端子排单元以扁平线缆连接。如果强行拉扯，可能导致断线。
请取出显示单元，注意避免强行拉扯。

4

主机安装接地请按照 D 种接地标准（接地电阻 100Ω以下）安装接地。
关于接地，请参阅“4-6 接地端子的连接”。

壳体的突起部分

壳体的突起部分

显示单元

接地螺栓

59 / 1714. 安装方法

54-7 线缆的连接将线缆防水接头或导管安装到主机的连接口线缆防水接头或导管应适合用途，且适合所使用的线缆种类。

电缆防水接头

电缆防水接头

6

拉出并取下端子排电路板上的端子排7将电源线缆和信号线缆连接到电源/信号用端子排请将各条线缆连接到电源/信号用端子排的以下端子上。接点用
线缆的连接请确认“4-9 警报继电器的设定（选配件）”。将线缆连接到端子排的方法，请参阅“4-7-5 将线缆连接到端子排”。
＜使用 3 线线缆时＞
端子编号
电源/信号线缆的连接
1
电源（+）
2
公共端（电源(-)、信号(-)）
3信号（+）

电源／信号用端子排接点输出用端子排

压接端子

绝缘体部分

导体部分

推荐绝缘体部分长度：40 - 50mm

连接线缆时，请将线缆的屏蔽连接到主机内部的左上方或右上方的端子上。
否则在某些安装环境下可能会受到噪声的影响。在那种情况
下请拆除屏蔽。
8

1
2
3
4（数字为端子编号）＜使用 4 线线缆时＞
端子编号
电源/信号线缆的连接
1
电源(+)
2
电源(-)
3
信号(+)
4
信号(-)

护套部分

1
2
3
4

为接点输出用端子排连接所需数量的接点用线缆
将线缆连接到端子排的方法，请参阅“4-7-5 将线缆连接到端子排”。

※ 无接点输出（选配件）时，没有接点输出用端子排。

60 / 1714. 安装方法

94-7 线缆的连接

将端子排安装到端子排电路板请调整线缆的位置，以避免向端子排施加负载。

电源／信号用端子排接点输出用端子排10 使显示单元返回原位置，用 2 个螺钉(M3)固定此时，请注意避免夹到线缆。

11 顺时针转动安装盖子，用 1 个内六角固定螺钉(M4)进行固定
请使用六角棒扳手（对边 2）等工具紧固内六角固定螺钉(M4)。
安装盖子时请注意避免夹到线缆。请切实安装盖子，直至顺时针无法转动。

※ 图为 SD-3。SD-3SC 的连接方法也相同。

61 / 1714. 安装方法

4-7 线缆的连接4-7-3 将电源线缆和信号线缆连接到遥控传感器头(GD-3)
1

旋松遥控传感器头(GD-3)盖子的 1 个内六角固定螺钉(M4)请使用六角棒扳手（对边 2）等工具旋松内六角固定螺钉(M4)。
旋松并取下盖子时，请注意避免盖子掉落。

2

逆时针转动盖子将其取下

3

将线缆防水接头或导管安装到遥控传感器头(GD-3)的连接口线缆防水接头或导管应适合用途，且适合所使用的线缆种类。
电缆防水接头

4

六角棒扳手

端子排

拉出并取下端子排电路板上的端子排(TN1)

端子排
端子排

62 / 1714. 安装方法

54-7 线缆的连接将电源线缆和信号线缆连接到端子排请将各条线缆连接到端子排的以下端子上。将线缆连接到端
子排的方法，请参阅“4-7-5 将线缆连接到端子排”
。
端子编号
1
2
3
4

电源/信号线缆的连接
电源(+)
电源(-)
信号(+)
信号(-)

4 3 21（数字为端子编号）

护套部分

绝缘体部分

导体部分

推荐绝缘体部分长度：50 - 60mm

连接线缆时，请将线缆的屏蔽连接到主机内部的左上方或右上方的端子上。
否则在某些安装环境下可能会受到噪声的影响。在那种情况
下请拆除屏蔽。
6

使线缆从主机内侧穿过连接口请让线缆防水接头或导管穿过安装侧的连接口。

7

将端子排安装到端子排电路板请调整线缆的位置，以避免向端子排施加负载。
电缆防水接头

＜线缆（外部导线）穿过右侧时＞＜线缆（外部导线）穿过左侧时＞

63 / 1714. 安装方法

84-7 线缆的连接顺时针转动安装盖子，用 1 个内六角固定螺钉(M4)进行固定
请使用六角棒扳手（对边 2）等工具紧固内六角固定螺钉(M4)。
六角棒扳手

安装盖子时请注意避免夹到线缆。请切实安装盖子，直至顺时针无法转动。

64 / 1714. 安装方法

4-7 线缆的连接4-7-4 连接主机(SD-3SC)和遥控传感器头(GD-3)
1

旋松主机(SD-3SC)盖子的 1 个内六角固定螺钉(M4)，然后逆时针转动盖子将其取下
请使用六角棒扳手（对边 2）等工具旋松内六角固定螺钉(M4)。
旋松并取下盖子时，请注意避免盖子掉落。

2

旋松 2 个固定显示单元的螺钉(M3)3取出显示单元，挂在壳体的突起部分
显示单元和端子排单元以扁平线缆连接。如果强行拉扯，可能导致断线。
请取出显示单元，注意避免强行拉扯。

壳体的突起部分

壳体的突起部分显示单元

4

将线缆防水接头或导管安装到主机的连接口线缆防水接头或导管应适合用途，且适合所使用的线缆种类。

电缆防水接头

65 / 1714. 安装方法

4-7 线缆的连接5

拉出并取下端子排电路板上的遥控连接用端子排6

旋松 1 个固定夹片的螺钉(M3)，取下夹片螺钉(M3)

7

夹片

将来自遥控传感器头(GD-3)的电源线缆和信号线缆连接遥控连接
用端子排请将各条线缆连接到遥控连接用端子排的以下端子上。将线缆连
接到端子排的方法，请参阅“4-7-5 将线缆连接到端子排”
。
端子编号
1
2
3
4

护套部分

电源/信号线缆的连接
电源（+）
电源（-）
信号（+）
信号（-）

屏蔽部分

绝缘体部分

导体部分
1

推荐屏蔽部分长度：30 - 40mm
推荐绝缘体部分长度：130 - 140mm

2

3

4

（数字为端子编号）遥控连接用端子排8

将端子排安装到端子排电路板请调整线缆的位置，以避免向端子排施加负载。

66 / 1714. 安装方法

94-7 线缆的连接对线缆使用屏蔽夹片请用 1 个螺钉(M3)进行固定，用夹片夹住线缆的
屏蔽部分。

屏蔽部分

螺钉(M3)

夹片

10 使显示单元返回原位置，用 2 个螺钉(M3)固定此时，请注意避免夹到线缆。

11 顺时针转动安装盖子，用 1 个内六角固定螺钉(M4)进行固定
请使用六角棒扳手（对边 2）等工具紧固内六角固定螺钉(M4)。
安装盖子时请注意避免夹到线缆。请切实安装盖子，直至顺时针无法转动。

67 / 1714. 安装方法

4-7 线缆的连接＜SD-3SC 和 GD-3 的连接示例＞
SD-3SC电缆防水接头

GD-3

线缆

＜SD-3/SD-3SC 和 RM-5003 的端子排连接示例（一例）＞
SD-3/SD-3SC

RM-5003

屏蔽线缆

1

13

2

12

3

14

4

68 / 1714. 安装方法

4-7 线缆的连接＜与指示计的连接示例＞＜SD-3＞

SD-3
将屏蔽连接到外壳的接地螺栓

指示计（RM-5003 等）

电源(+)
公共端
信号(+)

＜SD-3SC＞
指示计（RM-5003 等）

SD-3SC将屏蔽连接到外壳的接地螺栓

电源(+)
公共端
信号(+)

夹住固定屏蔽线缆
GD-3
将屏蔽连接到外壳的接地螺栓

69 / 1714. 安装方法

4-7 线缆的连接4-7-5 将线缆连接到端子排将线缆剥到规定长度，插入端子排的电线插入口。

<端子排的规格> 额定电压：AC 300V
 额定电流：10A
但是，因所使用的线缆而异。

＜连接条件＞
 使用线缆：0.25mm2 - 2.5mm2
 最大包覆外径：φ4.1mm
 剥皮线（导体部分）的长度：8 - 9mm（仅 SD-3 TN1，9 - 10mm）
 连接工具：一字螺丝刀（刃口 2.5 × 0.5mm）或操作杆（附件）注意请务必遵守规定的剥皮线（导体部分）长度剥除线缆外皮。
• 剥皮线（导体部分）长度较短，未能正确夹紧电线时，可能导致通电不良或发热。
• 剥皮线（导体部分）长度较短，咬住电线外皮时，可能导致通电不良或发热。
• 剥皮线（导体部分）长度较长，电线外露时，可能导致绝缘不良或短路。
• 请注意电线散股。如果插入时电线散股，可能导致绝缘不良或发热。
• 为预防上述情况，如果剥皮线是绞线，建议使用棒状端子（环箍）。
9 - 10mm＜适合的棒状端子＞
使用棒状端子时，请使用以下产品。
 棒状端子（环箍）：型号 216 系列（WAGO 产） 压接工具：型号压线钳 4 (206-1204)（WAGO 产）注意• 棒状端子请务必使用指定产品。使用非指定的棒状端子时无法保证性能。

70 / 1714. 安装方法

4-7 线缆的连接＜将线缆连接到端子排＞为了将线缆连接到端子排，请使用标准附件操作杆或一字螺丝刀（刃口 2.5 × 0.5mm）。
将电源及信号线缆连接到 SD-3 的 TN1 时，不可以使用标准附件操作杆。请使用一字螺丝刀。

注意• 请务必使用正确的工具。
• 原则上 1 个电线插入口只能连接 1 条线缆。
• 将线缆插入螺丝刀插槽后，由于接触不到导电部，可能导致通电不良或发热。

使用标准附件操作杆时，如下所示，将线缆连接到端子排。
1

如右图所示，将操作杆挂到端子排的螺丝刀插槽（方孔）

2用手指按操作杆，按下端子排中间的弹簧

螺丝刀插槽
电线插入口

操作杆

3

按着操作杆，将线缆插入电线插入口（圆孔），直到顶住为止4放开操作杆线缆连接到端子排。
请轻拉线缆，以确认是否已正确连接。
请勿强力拉扯。

71 / 171线缆

4. 安装方法

4-7 线缆的连接在连接到 SD-3 TN1 的过程中使用一字螺丝刀时，如下所示，将线缆连接到端子排。
1

如右图所示，将一字螺丝刀插入端子排的螺丝刀插槽
（方孔）

一字螺丝刀

螺丝刀插槽
电线插入口

2

在插入一字螺丝刀的状态下，将线缆插入电线插入口（圆孔），直到顶住为止

线缆

3

放开一字螺丝刀
线缆连接到端子排。
请轻拉线缆，以确认是否已正确连接。
请勿强力拉扯。

72 / 1714. 安装方法

4-8 安装传感器

4-8 安装传感器将传感器安装到主机(SD-3)或遥控传感器头(GD-3)。
1

如果已安装抽吸帽或防溅罩、过滤器，将其拆下2

旋松固定传感器护板的 1 个内六角固定螺钉(M4)请使用六角棒扳手（对边 2）等工具旋松内六角固定螺钉
(M4)。

传感器护板
六角棒扳手

3

转动并取下传感器护板
请使用六角棒扳手（对边 8）或活动扳手将其取下。

4

安装有传感器时，将其拉出取下

传感器
传感器护板
六角棒扳手

5

对齐主机侧连接器的方向和传感器侧连接器的方向，
将传感器插到深处请将传感器切实插入主机侧连接器的深处。
插入传感器时，请注意避免强行转动或强行按压，否则可能导致主机损坏。

主机侧连接器73 / 1714. 安装方法

6

4-9 警报继电器的设定（选配件）确认传感器已切实安装到深处

传感器

传感器未安
装到深处

7

将传感器护板安装到主机
请使用六角棒扳手（对边 8）或活动扳手进行安装。

8

用 1 个内六角固定螺钉(M4)固定传感器护板
请使用六角棒扳手等工具紧固内六角固定螺钉(M4)。

传感器护板※ 图为 SD-3。遥控传感器头(GD-3)的安装方法也相同。

注意• 更换传感器时，某些传感器如有异物混入，可能导致指示值波动。

4-9 警报继电器的设定（选配件）设定警报继电器后，可将本仪器作为独立单元进行操作。
警报继电器在默认状态下设定为励磁。

警告• 使用 AC 30V 或 DC 42.4V 以上的电压时，请将保护管安装到线缆上，或使用双层绝缘线缆。

74 / 171六角棒扳手

4. 安装方法

4-9 警报继电器的设定（选配件）注意• 在常时闭合的状态下使用警报继电器时，微小的振动等也可能引发接点部分烧熔。
• 为本仪器接通电源前，请连接警报继电器的线缆和传感器。

为了设定警报继电器，将警报器的配线连接到主机(SD-3、SD-3SC)的 TN2(Relay1)、TN3(Relay2)、TN4(Relay3)
的端子排。
1

旋松主机盖子的 1 个内六角固定螺钉 (M4)，然后逆时针转动盖子将其取下请使用六角棒扳手（对边 2）等工具旋松内六角固定螺钉
(M4)。
旋松并取下盖子时，请注意避免盖子掉落。

2

旋松 2 个固定显示单元的螺钉(M3)3取出显示单元，挂在壳体的突起部分
显示单元和端子排单元以扁平线缆连接。
如果强行拉扯，可能导致断线。
请取出显示单元，注意避免强行拉扯。

壳体的突起部分

壳体的突起部分显示单元

75 / 1714. 安装方法

454-9 警报继电器的设定（选配件）拉出并取下端子排电路板上的端子排(TN2(Relay1)、
TN3(Relay2)、TN4(Relay3))

TN2

将警报器的配线连接到已拆下端子排的对应端子
使用端子排的端子 1(N.O.)或端子 3(N.C.)的任意一个。
将线缆连接到端子排的方法，请参阅“4-7-5 将线缆连接到端子排”。
TN2(Relay 1)、TN3(Relay2)、TN4(Relay3)
通用（默认设定时（正常时非励磁）的端子分配）
端子编号
1
2
3

线缆的连接
N.O.
公共端
N.C.

TN4

N.O.：NormAl Open
N.C.：NormAl Close

护套部分

TN3

绝缘体部分

导体部分

推荐绝缘体部分长度：120 - 130mm

6

将端子排(TN2(Relay1)、TN3(Relay2)、TN4(Relay3))
安装到端子排电路板请调整线缆的位置，以避免向端子排施加负载。

7

使显示单元返回原位置，用 2 个螺钉(M3)固定此时，请注意避免夹到线缆。

8顺时针转动安装盖子，用 1 个内六角固定螺钉(M4)进行
固定
请使用六角棒扳手（对边 2）等工具紧固内六角固定螺钉(M4)。
安装盖子时请注意避免夹到线缆。请切实安装盖子，直至顺时针无法转动。

※ 图为 SD-3。SD-3SC 的连接方法也相同。

76 / 1714. 安装方法

4-10 安装防溅罩4-10 安装防溅罩
使用除硅过滤器或活性碳过滤器时，请先将过滤器装入防溅罩，再安装防溅罩。
1

使用除硅过滤器或活性碳过滤器时，请拆下防溅罩的盖子，将过滤器装入防溅罩后装上盖子
不使用除硅过滤器或活性碳过滤器时，不需要本步骤。

盖子

过滤器

2

为主机的护罩套上防溅罩，顺时针转动直至固定
请切实将防溅罩安装到位，直至顺时针无法转动。
此时，请注意避免转得太紧。

护板

防溅罩

※ 图为 SD-3。遥控传感器头(GD-3)的安装方法也相同。

77 / 1714. 安装方法

4-10 安装防溅罩

注记 防溅罩有 3 种：IRF 传感器防溅罩、可燃性传感器防溅罩、ESF 传感器防溅罩。
各种防溅罩的底面刻有识别编号。
·IRF 传感器防溅罩识别编号：1·NCF/SGF/SHF 传感器防溅罩识别编号：2
识别编号·ESF 传感器防溅罩识别编号：3
 可以使用各种防溅罩的 SD-3、GD-3 的型号如下。
＜IRF 传感器用>＜NCF/SGF/SHF 传感器用>＜ESF 传感器用>
SD-3RI
SD-3NC
SD-3ECGD-3RI
SD-3GH
SD-3ECB
SD-3GHSGD-3EC
SD-3SP
GD-3ECB
GD-3NCGD-3GH
GD-3GHS
GD-3SP

78 / 171

5. 使用方法

5-1 启动准备

5

使用方法
5-1 启动准备
连接电源前，请完成以下确认工作。如果不完成这些确认，可能导致触电或损坏仪器。
 请确认本仪器已接地。
 请确认已正确完成与外部之间的布线。
 请确认供给电源电压在额定值内。
 调整中外部接点可能会运行。如果接点已运行，请采取措施，使接点即使运行也不影响外部。

＜如何查看显示例＞
在本使用说明书中，LED 的显示例表示以下含义。
此外，显示例中检测对象气体为可燃性气体（单位：%LEL）。
PW

AL1 AL2 F

PPM %LEL%PW：POWER（电源指示灯）

AL1：ALM1（警报灯）点亮（红色）

AL2：ALM2（警报灯）点亮（红色）F：FAULT（故障灯）

点亮（黄色）

PPM：单位指示灯点亮（绿色）%LEL ：单位指示灯

点亮（绿色）%

点亮（绿色）：单位指示灯79 / 171

点亮（绿色）/

闪烁（绿色）

5. 使用方法

5-2 启动

5-2 启动

注意• 给本仪器接通电源(DC 24V)之前，请确认本仪器有无正确安装。

接通电源(DC 24V)，启动本仪器。
接通电源后约 25 秒，执行仪器的系统确认和警报断开等初始清空操作，进入检测模式。
接通电源后，指示灯自动切换如下。
通常接通电源后，在检测模式下使用。

RI

NC

EC

指示灯全部点亮GH

初始显示（约 25 秒）

检测模式

SP

指示灯全部点亮

初始显示
（约 25 秒）

初始显示

检测模式

（数字表示暖机时间）注意• 初始清空中绝对不要关闭电源。初始清空中对内部存储器进行读取。
• 启动后或更换传感器时，还需要根据传感器的种类进行相应的暖机运行。请按照规定时间进行暖机运行。
半导体式传感器(SGF)、热线型半导体式(SHF)在初始清空后显示暖机完成参考时间。此外，在暖机运行期间，警报运行和输出信号会变得不稳定。请事先通知相关部门，以避免发生异常。
• 暖机运行结束后，请进行气体校正。

80 / 171

5. 使用方法

5-3 基本运行流程5-3 基本运行流程电源接通后，按下 MENU/ESC 键，本仪器进行如下运行。
＜检测模式＞
MENU/ESC 键（长按）复位后
重新接通电源MENU/ESC 键（长按）＜第一警报＞＜用户模式＞＜第二警报＞＜维护模式＞＜故障警报＞

警告• 处于警报状态时，如果从检测模式切换到用户模式或维护模式，警报即被解除。

注记 本仪器的 LED 为 7 段数码管 5 位显示，因此按照项目编号显示设定内容。
在用户模式或维护模式下进行设定时，请选择与设定内容对应的项目编号进行设定。
（参阅“5-5-1 用户模
式的设定项目”、“7-2-1 维护模式的设定项目”）

81 / 171

5. 使用方法

5-4 操作方法

5-4 操作方法
本仪器使用控制钥匙进行操作。
使用控制钥匙，通过触控本仪器正面的 4 个键（MENU/ESC 键、SET 键、▲键、▼键）进行操作。
在本使用说明书中，使用控制钥匙触控各键的操作称为“按下”。
＜控制钥匙＞

注意• 请勿将控制钥匙强力按压在玻璃表面上。

82 / 171

5. 使用方法

5-5 用户模式

5-5 用户模式
切换到用户模式，即可进行零位校正和确认各种设定内容。

注意• 请勿随意变更设定。如果不了解设定内容随意变更设定，可能导致无法正确运行。

5-5-1 用户模式的设定项目
LED 显示

项目编号
1-0

设定项目
版本确认

1-1

零位校正

进行零位校正。
（参阅“5-6 进行零位校正”）

1-2

设定值显示

显示各种设定值。
（参阅“5-7 确认设定值”）

1-3切换到维护模式切换至维护模式。
（参阅“7-2 维护模式”）

83 / 171

内容
显示程序的版本等。
客户方通常不使用。

5. 使用方法

5-5 用户模式

5-5-2 切换到用户模式
1

在检测模式下长按 MENU/ESC 键（约 3 秒）

切换到用户模式。

2按下▲键或▼键，选择项目编号，再按下 SET 键

显示例：选择显示设定值时会显示所选择的项目编号的内容。
在各项目中进行设定或显示。

3操作结束后，长按 MENU/ESC 键（约 3 秒）
切换到检测模式。

警告• 操作结束后，请务必长按 MENU/ESC 键返回检测模式。本仪器在调整/设定状态下不检测气体。如果在用
户模式下放置不管，将在 10 小时后返回检测模式。其间变为无监测状态，非常危险。

84 / 171

5. 使用方法

5-6 进行零位校正

5-6 进行零位校正为了正确测量气体浓度，进行必要的零位校正。
进行零位校正之前，请准备零位校正用的气体，并连接到本仪器。
关于零位校正的准备，请参阅“7-4-1 气体校正的准备”。

警告• 使用周围空气进行零位校正时，请先确认周围是清洁的大气。如果在有杂质气体的状态下进行零位校正，将无法准确校正，实际发生气体泄漏时很危险。

注意• 请供应零位校正用的气体，待指示值稳定后再进行零位校正。

1

在用户模式下按下▲键或▼键，选择[1-1]，再按下 SET 键

2

导入零位校正用的气体，按下 SET 键

零位校正开始。

3

确认零位校正的结果
零位校正成功时
将会显示[PASS]，3 秒后自动返回显示[1-1]。

零位校正失败时
将会显示[FAIL]，3 秒后自动返回显示[1-1]。

85 / 171

5. 使用方法

5-6 进行零位校正

注记 由于零点（氧气则为 N2）在零附近大幅变动等原因，零位校正失败时，显示[FAIL]后返回显示[1-1]。此时
零位校正未完成。
 在缺氧警报规格中，[1-1]表示需进行间距校正。请导入清洁的大气，调整为[20.9](vol%)。关于间距校正，请参阅“7-4-3 进行间距校正”。

86 / 171

5. 使用方法

5-7 确认设定值

5-7 确认设定值显示各种设定值。
项目编号
1-2
1-2.0

设定项目
设定值显示
警报设定显示

1-2.1

-

备注
零位跟踪设定 ON/OFF 显示：

NC灵敏度校正 ON/OFF 显示

：

EC

EC

客户方通常不使用。

5-7-1 确认警报设定值
显示第一警报点、第二警报点、警报延迟时间等警报设定。

注记 零位跟踪设定 ON/OFF 显示仅在检测原理为新型陶瓷式和恒电位电解式时显示。
（参阅“1-2-2 各型号的检测原理和检测对象气体”） 灵敏度校正 ON/OFF 显示仅在检测原理为恒电位电解式时显示。
（参阅“1-2-2 各型号的检测原理和检测
对象气体”
）

1

在用户模式下按下▲键或▼键，选择[1-2]，再按下 SET 键

2选择[1-2.0]，按下 SET 键

3确认警报设定的各设定值
按下▲键或▼键，切换设定值的显示。
· 第一警报点显示
显示第一警报点。

87 / 171

5. 使用方法

5-7 确认设定值· 第二警报点显示
显示第二警报点。

· 警报延迟显示
显示警报延迟时间（秒）。

· 零抑制值显示
显示正侧的抑制值。

· 零位跟踪 ON/OFF 显示
显示零位跟踪设定的 ON/OFF。
（通常客户不能变更设定。
）· 灵敏度校正 ON/OFF 显示显示灵敏度校正的 ON/OFF。
（通常客户不能变更设定。
）

按下▲键，即返回第一警报点显示。
第一警报点显示
4

按下 MENU/ESC 键
返回显示[1-2.0]。

注记 警报设定值的警报点、警报延迟时间、零抑制值可以在维护模式的环境设定 1 中进行设定。
（参阅“7-5
进行环境设定 1”）

88 / 171

5. 使用方法

5-8 结束5-8 结束如需结束本仪器的运行，请停止检测模式状态下的电源(DC 24V)供给。

警告• 结束本仪器的运行后，上位（中央）系统可能发出警报。
• 如需结束本仪器的运行，请先在上位（中央）系统中将 INHIBIT 设定为 ON（停止警报处理）后再结束运行。另外，请确认本仪器的外部输出、外部触点输出端子上连接着的仪器的动作情况，判断是否可以切断电源。
• 使用接点时（选配件），如果停止电源供给，接点可能会运行。

89 / 1716. 警报动作

6-1 气体警报运行6警报动作
6-1 气体警报运行
6-1-1 警报运行
检测到的气体浓度或氧气浓度达到或超过警报设定值时，本仪器的警报会运行。
（自动复位或自我保持）注记 警报设定值（第一警报、第二警报）和警报运行（自我保持、自动复位）在出厂时已预先设定好。
 为防止误动作，设定了警报延迟时间（初始设定：2 秒）。如果不需要，请将其解除。

＜气体浓度显示＞
超过检测范围（满量程），LED 显示即变为“∩∩∩∩”。

＜电源/警报灯显示＞
警报变为 2 级警报。各自达到或超过警报设定值时点亮。
警报灯(ALM1)、警报灯(ALM2)点亮（红色），显示如下。
电源指示灯(POWER)保持连续点亮（绿色）。
＜第一警报＞＜第二警报＞
警报灯(ALM1)
警报灯(ALM2)

警报灯(ALM1)

ALM1 点亮（红色）ALM1 和 ALM2 点亮（红色）注记 警报运行为自我保持时，即使气体浓度低于警报设定值，也会持续发出警报。
请在确认周围没有气体后进行按键操作。通过按键操作解除警报状态。

90 / 1716. 警报动作

6-1 气体警报运行

6-1-2 接点运行（自动复位）如果设定为自动复位，气体浓度达到或超过警报设定值时接点运行。
在气体浓度低于警报设定值时，触点动作自动复位。

＜警报模式(H-HH)＞
正 常

警 报

复 位

气体浓度
第二警报设定值(ALM2)
第一警报设定值(ALM1)■ ALM1 警报灯（红）■ ALM2 警报灯（红）■ ALM1 警报接点
■ ALM2 警报接点

＜警报模式(L-H)＞
正 常

警 报

复 位

警 报

复 位

气体浓度
第二警报设定值(ALM2)
第一警报设定值(ALM1)■ ALM1 警报灯（红）■ ALM2 警报灯（红）■ ALM1 警报接点
■ ALM2 警报接点

＜警报模式(L-LL)＞（缺氧警报）
正 常

警 报

气体浓度
第一警报设定值(ALM1)
第二警报设定值(ALM2)■ ALM1 警报灯（红）■ ALM2 警报灯（红）■ ALM1 警报接点
■ ALM2 警报接点

91 / 171

复 位

6. 警报动作

6-1 气体警报运行

6-1-3 接点运行（自我保持）如果设定为自我保持，气体浓度达到或超过警报设定值时接点运行。
重置后如果气体浓度低于警报设定值，接点运行自动复位。
警报灯在警报发出时闪烁。重置后警报灯变为点亮，气体浓度低于警报设定值时熄灭。

注记 在检测模式下，按下 MENU/ESC 键、SET 键、▲键、▼键的任意键，即可解除警报运行。

＜警报模式(H-HH)＞
正 常

警 报

复 位

气体浓度
第二警报设定值(ALM2)
第一警报设定值(ALM1)
重置

重置■ ALM1 警报灯（红）■ ALM2 警报灯（红）■ ALM1 警报接点
■ ALM2 警报接点
重置

■ ALM1 警报灯（红）■ ALM2 警报灯（红）■ ALM1 警报接点
■ ALM2 警报接点
重置

■ ALM1 警报灯（红）■ ALM2 警报灯（红）■ ALM1 警报接点
■ ALM2 警报接点
重置

■ ALM1 警报灯（红）■ ALM2 警报灯（红）■ ALM1 警报接点
■ ALM2 警报接点

92 / 1716. 警报动作

6-1 气体警报运行＜警报模式(L-H)＞
正 常

警 报

复 位

警 报

复 位

气体浓度
第二警报设定值(ALM2)
第一警报设定值(ALM1)
重置

重置■ ALM1 警报灯（红）■ ALM2 警报灯（红）■ ALM1 警报接点
■ ALM2 警报接点
重置

重置

■ ALM1 警报灯（红）■ ALM2 警报灯（红）■ ALM1 警报接点
■ ALM2 警报接点
重置

■ ALM1 警报灯（红）■ ALM2 警报灯（红）■ ALM1 警报接点
■ ALM2 警报接点
重置

■ ALM1 警报灯（红）■ ALM2 警报灯（红）■ ALM1 警报接点
■ ALM2 警报接点

93 / 1716. 警报动作

6-1 气体警报运行＜警报模式(L-LL)＞（缺氧警报）
正 常

警 报

复 位

气体浓度
第一警报设定值(ALM1)
第二警报设定值(ALM2)
重置

重置■ ALM1 警报灯（红）■ ALM2 警报灯（红）■ ALM1 警报接点
■ ALM2 警报接点
重置

■ ALM1 警报灯（红）■ ALM2 警报灯（红）■ ALM1 警报接点
■ ALM2 警报接点
重置

■ ALM1 警报灯（红）■ ALM2 警报灯（红）■ ALM1 警报接点
■ ALM2 警报接点
重置

■ ALM1 警报灯（红）■ ALM2 警报灯（红）■ ALM1 警报接点
■ ALM2 警报接点

94 / 1716. 警报动作

6-2 故障警报运行

6-1-4 发出警报时的应对措施
警报发出后，请按照客户的管理规则及时应对。
通常应进行以下应对。
Step1 确认本仪器的指示值。

注意• 如为瞬间的气体泄漏，确认时指示值可能已经降低。除气体警报以外，因噪声或偶发条件触发临时的警报状态时，指示值也可能会降低。

Step2 根据警报管理浓度从监控区域疏散人员，确保安全。
Step3 如果浓度显示持续，应关闭问题气体的总阀，确认气体浓度已经降低。
Step4 前往气体泄漏现场，使用便携式气体检测仪等确认气体的残留情况。

注意• 请假设有气体残留，佩戴好装备以避免危险，在此基础上进行作业。
Step5 确认没有危险后，对气体泄漏进行处理。

6-2 故障警报运行
如果本仪器内检测到异常运行，就会发出故障警报。
发出故障警报后，故障灯(FAULT)点亮（黄色），显示如下的错误编号。
＜传感器异常时＞
故障灯(FAULT)FAULT 点亮（黄色）注记 从故障状态恢复正常时，执行初始清空并重新启动。
 关于故障内容（错误编号）及其处理方法，请参阅“9 故障排除”。
 发出故障警报时，请查明原因，正确进行处理。仪器有问题、故障频发时，请及时联系本公司。

95 / 1716. 警报动作

6-3 警告运行6-3 警告运行
如果本仪器的使用环境温度超过规格范围，浓度显示和警告编号“E-27”会交替显示。
此时并非本仪器的故障，请继续进行气体测量。外部输出值也根据测量气体浓度输出 4～20mA，故障灯也不运行。

96 / 1716. 警报动作

6-4 外部输出运行

6-4 外部输出运行
6-4-1 测量气体非氧气时
4 - 20mA（输出值）

模式
1

电源断开

0mA

2

初始清空

2.5mA 设定：2.5mA
4mA、HOLD、4 - 20mA 设定：4mA

3

检测模式（无警报时）4 - 20mA（浓度输出）

4

检测模式（气体警报时）4 - 20mA（浓度输出）

5

检测模式（满量程）20.5mA（固定）6

使用温度范围外

4 - 20mA（浓度输出）

7

故障警报

0.5mA（固定）8

INHIBIT

2.5mA 设定：2.5mA
4mA、HOLD、4 - 20mA 设定：4mA

9警报测试模式输出 ON 设定：4～20mA（浓度输出）
输出 OFF 设定：4mA

10

用户模式

2.5mA 设定：2.5mA
4mA 设定：4mA
HOLD 设定：保留前值
4 - 20mA 设定：4 - 20mA（浓度输出）

11

维护模式

2.5mA 设定：2.5mA
4mA 设定：4mA
HOLD 设定：保留前值
4 - 20mA 设定：4 - 20mA（浓度输出）

97 / 1716. 警报动作

6-4 外部输出运行＜气体浓度和外部输出示例（为标准量程时）＞＜4 - 20mA 规格（维护输出：2.5mA 设定）＞
检测模式
维护模式

外部输出
20.5mA
20mA

4mA
2.5mA
气体浓度
0 零抑制

全量程
20mA 输出

满量程
20.5mA 固定输出注意
• 4 - 20mA 已经调整完毕。为满量程时，不进行约 20.5mA 以上的输出。
• INHIBIT 功能处于 ON 时或初始清空中时，输出取决于维护模式中的 4 - 20mA 输出设定。本仪器启动时或规格变更时请特别注意。
• 请充分理解运行内容，视需要采取措施，以避免接收侧发出误警报。
• 禁用抑制功能时，信号将变为 4mA (0%LEL)的线性输出。
• 禁用负侧的抑制功能时，信号将变为 3.2mA (-5%LEL)的线性输出。

98 / 1716. 警报动作

6-4 外部输出运行＜气体浓度和外部输出示例（为双量程时）＞ NC如果检测原理为新型陶瓷式，则具备 L 量程和 H 量程的 2 个指示范围。
指示的可燃性气体浓度超过 L 量程的全量程时，自动切换到 H 量程。另外，气体浓度降低到 L 量程的全量程以下
时，再次切换到 L 量程。
例如，如果指示对象气体为异丁烷，则指示范围 0 - 2000ppm 处于 L 量程，变为 12.0 - 100.0%LEL 时则切换到
H 量程。
＜4 - 16mA（L 量程）/16 - 20mA（H 量程）设定＞
外部输出
20.5mA
20mA

检测模式
维护模式

16mA

4mA
2.5mA
L 量程

0 零抑制

H 量程
全量程
20mA 输出

气体浓度
满量程
20.5mA 固定输出＜4 - 20mA（L 量程）/22mA（H 量程）设定＞

外部输出
20.5mA
20mA

检测模式
维护模式

4mA
2.5mA
L 量程

0

全量程
20mA 输出

零抑制

99 / 171

气体浓度
H 量程
20.5mA 固定输出6. 警报动作

6-4 外部输出运行＜4 - 20mA（H 量程）设定＞
外部输出

检测模式
维护模式

20.5mA20mA

切换点当量 mA
满量程
20.5mA 固定输出

4mA
2.5mA
L 量程

H 量程
气体浓度

0 零抑制

全量程
20mA 输出

100 / 171

6. 警报动作

6-4 外部输出运行

6-4-2 测量气体为氧气时
4 - 20mA（输出值）

模式
1

电源断开

0mA

2

初始清空

2.5mA 设定：2.5mA
4mA、HOLD、4 - 20mA 设定：4mA※

3

检测模式（无警报时）4 - 20mA（浓度输出）

4

检测模式（气体警报时）4 - 20mA（浓度输出）

5

检测模式（满量程）20.5mA（固定）6

使用温度范围外

4 - 20mA（浓度输出）

7

故障警报

0.5mA（固定）8

INHIBIT

2.5mA 设定：2.5mA
4mA、HOLD、4 - 20mA 设定：4mA※

9警报测试模式输出 ON 设定：4～20mA（浓度输出）
输出 OFF 设定：保留前值

10

用户模式

2.5mA 设定：2.5mA
4mA 设定：4mA※
HOLD 设定：保留前值
4 - 20mA 设定：4 - 20mA（浓度输出）

11

维护模式

2.5mA 设定：2.5mA
4mA 设定：4mA※
HOLD 设定：保留前值
4 - 20mA 设定：4 - 20mA（浓度输出）※ESF（检测对象气体：氧气/0 - 25vol%）为 Air 当量(20.9vol% = 17.4mA)

101 / 171

6. 警报动作

6-4 外部输出运行

注意＜模拟传输(4 - 20mA)时＞• 4 - 20mA 已经调整完毕。为满量程时，不进行约 20.5mA 以上的输出。
• INHIBIT 功能处于 ON 时或初始清空中时，输出取决于维护模式中的 4 - 20mA 输出设定。使用 ESF（检测对象气体：氧气）时，请注意在初始清空中可能有 2.5mA 等的低输出。本仪器启动时或规格变更时请特别注意。
• 请充分理解运行内容，视需要采取措施，以避免接收侧发出误警报（设为跳过状态等）。
• 特别在以下情况下需注意。
模拟传输
需注意的事例
4 - 20mA 规格
在维护模式下，或 INHIBIT 功能处于 ON 时，
使用 ESF（检测对象气体：氧气）(L-LL､ L-H)
输出 2.5mA 时，上位（L 警报）系统可能会发出
误警报。
规格变更
从其他原理(H-HH)变更过来时，在变更确定前（显变更为 ESF（检测对象气体：氧气/0 - 25vol%）示［C-02］前），将会输出 4mA（相当于零浓度）。
规格变更从 0 - 25vol%(L-LL、L-H)变更过来时，在变更确从 ESF（检测对象气体：氧气/0 - 25vol%）变更
定前（显示[C-02]前），将会输出 17.4mA（相当过来于约 84%F.S.）。

102 / 171

6. 警报动作

6-4 外部输出运行＜气体浓度与外部输出示例＞
0 - 25vol%（缺氧警报规格）
外部输出
20.5mA
20mA
检测模式
维护模式

17.4mA

4mA

气体浓度
0
抑制值

全量程
20mA 输出

满量程
20.5mA 固定输出注意
• 4 - 20mA 已经调整完毕。为满量程时，不进行约 20.5mA 以上的输出。
• INHIBIT 功能处于 ON 时或初始清空中时，输出取决于维护模式中的 4 - 20mA 输出设定。本仪器启动时或规格变更时请特别注意。
• 请充分理解运行内容，视需要采取措施，以避免接收侧发出误警报。
• 禁用负侧的抑制功能时，信号将变为 3.2mA (-5%LEL)的线性输出。

103 / 171

6. 警报动作

6-5 发出正确警报所需的各种功能

6-5 发出正确警报所需的各种功能
6-5-1 零抑制功能本仪器使用的传感器受环境变化的影响（温度特性、湿度特性等）及干扰气体的影响（干扰特性），会影响本仪器
的指示值。因此，即使仪器正常时，也有可能在零位附近看到指示值变动的情况。
零抑制功能是使零位附近的环境变化或干扰气体等的影响淡化的功能。使用本功能，可以隐藏（抑制）低于设定值
的指示变动，指示值变为零（氧气则为[20.9] (vol%)）。
＜无零抑制功能＞＜有零抑制功能＞

零位附近的变动显示为指示值在设定值的范围内显示为零注记
 零抑制功能的标准设定值因传感器规格而异。
 如果检测对象气体为氧气，启用零抑制功能后，20.9vol%附近的变动在设定值之前指示值均为[20.9](vol%)。检测对象气体为氧气时，零抑制功能的标准设定值为±0.5vol%（相对于 20.9vol%）。
 在用户模式和维护模式下本功能将被解除，发生低于设定值的指示值变动。
 零抑制方式在初始设定中设定为[CUT]。
 正侧的零抑制值在 0 - 第一警报点的 2/3（如果第一警报点为 25%LEL，则是 16%LEL 以内）的范围内可以变更。此外，负侧的零抑制值在全量程的 5% + 1digit（如果全量程为 100%LEL，则是-5%LEL+1digit）
的范围内可以变更。
（客户方通常不使用。
）

注意• 负侧的零抑制值（零位潜藏）被抑制全量程的 5%。如果指示值低于全量程的 5%，将会显示错误编号“E-1”。在此状态下不能进行准确的气体检测，因此请确认周围没有杂质气体，进行零位校正。

104 / 171

6. 警报动作

6-5 发出正确警报所需的各种功能

6-5-2 传感器自动识别功能本仪器具备在更换传感器后或变更传感器规格后自动识别传感器的功能。
如果安装了不同制造编号的传感器、不同工作原理或规格的传感器，将会显示以下信息以防止误装。

＜更换了传感器时＞
定期更换时，如果换上了相同规格的传感器（原理、型号等），将会显示以下信息。
按下 MENU/ESC 键，就会识别为新传感器并启动。

如果更换后的传感器的制造日期与更换前的传感器相同或更旧，将会交替显示如下的[C-01]和[USEd]。
可能误装了旧的传感器。请确认传感器。

＜变更了规格时＞
与原先安装的传感器相比较，如果换上了不同规格的传感器（原理、型号等），将会显示以下信息。
按下 MENU/ESC 键，就会识别为新规格的传感器并启动。
如果无意变更规格，则可能是误装。请确认传感器。

注意• 显示[C-02]时，如果确定传感器规格发生已变更（原理、传感器类型、检测气体、检测范围等），则气体检测仪的规格也将变更。
此时除警报设定值以外，以下所示的参数也会被变更为初始设定值。
如果要在初始设定值以外使用，请在维护模式下变更。
·警报延迟
·零抑制值
·警报方式
• 显示[C-02]，按下 MENU/ESC 键确定后，初始清空也将同时开始并切换运行。此外，初始清空输出也将
同步切换。
• 在维护模式下，外部输出的设定会保持变更前的设定。请特别注意。
• ESF 规格的警报方式有 3 种：L-LL、L-H、H-HH。
关于“从 ESF（L-LL 警报）开始变更”或“变更为 ESF（L-LL 警报）”，警报的方向相反。

105 / 171

7. 保养检查

7-1 检查的频率与检查项目

7保养检查本仪器是防灾、安全保障上重要的仪器。
为维护本仪器性能，提高防灾和安全保障上的可靠性，请定期实施维护、检查。

7-1 检查的频率与检查项目
使用本仪器前，请定期检查以下项目。
 日常检查
 月度检查
 定期检查

：作业前请进行检查。
：请 1 个月进行 1 次警报电路检查（警报测试）。
：请使用校正用气体确认检测精度和进行气体校正。
＜推荐气体校正周期＞ ESF/IRF 传感器应每 6～12 个月至少校正 1 次
NCF/SGF/SHF 传感器应每 4～6 个月至少校正 1 次
为保持作为安全仪器的性能，请按照每年至少 1 次的频率进行气体校正。
关于传感器保修条件，请确认本书末尾的保修规定。

注记 气体校正请按照先零位校正、后间距校正的顺序进行。
 仅 SGF 传感器是按照先间距校正、后零位校正的顺序进行。显示的值为零抑制值以上时，无法进行零位
校正。

检查项目

检查内容

日常
检查

1 个月
检查

定期
检查

确认电源

请确认电源指示灯已点亮。

〇

〇

〇

确认浓度显示请确认浓度显示值为零（氧气则为[20.9](vol%)）。
指示值有偏离时，请确认周围有无杂质气体，进行零位
校正。

〇

〇

〇

警报测试

请使用警报测试功能，进行警报电路检査。
如果使用警报继电器（选配件），请同时进行确认。

-

〇

〇

气体校正

请使用校正用气体进行灵敏度校正。

-

-

〇

气体警报确认

请使用校正用气体确认气体警报。

-

-

〇

接地线确认

为保持 EMC 性能，请确认接地线的连接没有松动或
腐蚀。

-

-

○

106 / 171

7. 保养检查

7-1 检查的频率与检查项目＜关于维护服务＞
本公司进行包含气体校正等在内的有关定期检查、调整、维护等服务。
制备校正用气体时，需要使用规定浓度的气瓶、气体袋等专用器具。
本公司指定的服务人员配备相关专用器具，具备与产品相关的专业知识。为了维持机器的安全动作，请使用本公司维护服务。
以下是维护服务的主要内容。详情请洽询本公司营业部。
服务

服务内容

确认电源

·确认电源电压。
·确认电源指示灯已点亮。
（确认在显示部可以识别电源指示灯(POWER)、警报灯(ALM1、ALM2)、故障灯(FAULT)、单位指示灯(PPM、%LEL、%)。）·使用安全电源时，确认使用安全电源的运行情况。

确认浓度显示使用零位校正气体确认浓度显示值为零。如指示值有偏差，进行零位校正。
在缺氧警报规格下导入清洁的大气，确认浓度显示值变为[20.9](vol%)。
如指示值有偏差，进行间距校正。

确认过滤器确认滤尘器的污垢情况及是否有堵塞。
如果有污垢或发生了堵塞，应更换滤尘器。

警报测试使用警报测试功能，进行警报的电路检査。
·警报灯确认（确认 ALM1、ALM2 各自的运行）·确认外部警报（确认蜂鸣器、复位信号等外部警报运行）

气体校正

使用校正用气体进行灵敏度校正。

气体警报确认

使用校正用气体确认气体警报。
·警报确认（确认达到警报设定值时是否发出警报）·延迟时间确认（确认发出警报之前的延迟时间）·警报灯确认（确认 ALM1、ALM2 各自的运行）·确认外部警报（确认蜂鸣器、复位信号等外部警报运行）

清扫、修缮仪器
（目视诊断）

确认仪器外观及罩盖、内部等的污垢及伤痕，清扫、修缮显眼的位置。
如有裂纹或破损，应更换部件。

仪器的操作确认

操作按键，确认各种功能的运行，进行参数等的检查。

更换劣化部件更换传感器、过滤器等劣化部件。

107 / 171

7. 保养检查

7-2 维护模式

7-2 维护模式
切换到维护模式后，除各种测试、零位校正、间距校正以外，还可以进行各种设定。

7-2-1 维护模式的设定项目
LED 显示

项目编号
2-0

设定项目
测试模式

2-1

零位校正

进行零位校正。
（参阅“7-4-2 进行零位校正”）

2-2

间距校正

进行间距校正。
（参阅“7-4-3 进行间距校正”）

2-3

显示

客户方通常不使用。

2-4

环境设定 1设定警报点、警报运行、故障运行、
零抑制值、日期时间等项目。
（参阅“7-5 进行环境设定 1”）

2-5

环境设定 2

客户方通常不使用。

2-6

环境设定 3

设定使用接点、各种外部输出等项目。
（参阅“7-6 进行环境设定 3”）

2-7-

客户方通常不使用。

108 / 171内容
进行气体测试、警报测试、故障测试、LED 测试和 RAM 测试。
（参阅“7-3 进行各种测试（测试模式）”）

7. 保养检查

7-2 维护模式

项目编号
2-8

设定项目
工厂模式切换

2-9

用户模式切换

LED 显示内容
客户方通常不使用。

切换到用户模式。
（参阅“5-5 用户模式”）

7-2-2 切换维护模式
1

在检测模式下长按 MENU/ESC 键（约 3 秒）

切换到用户模式。

2按下▲键或▼键，选择[1-3]，再按下 SET 键

切换到维护模式。

3按下▲键或▼键，选择项目编号，再按下 SET 键

显示例：选择环境设定 1 时会显示所选择的项目编号的内容。
在各项目中进行设定。

109 / 171

7. 保养检查

4

7-2 维护模式

操作结束后，长按 MENU/ESC 键（约 3 秒）
切换到检测模式。

警告• 操作结束后，请务必长按 MENU/ESC 键返回检测模式。本仪器在调整/设定状态下不检测气体。如果在维护模式下放置不管，将在 10 小时后返回检测模式。其间变为无监测状态，非常危险。

注记 如果密码设定为 ON，切换到维护模式时需输入密码。
 密码由本公司事先设定。
如有疑问，请联系经销商或就近的本公司营业网点。

＜从维护模式切换到用户模式＞
1

在维护模式下多次按下▲键或▼键，选择[2-9]，
再按下 SET 键

2

按下 SET 键

切换到用户模式。

110 / 171

7. 保养检查

7-3 进行各种测试（测试模式）

7-3 进行各种测试（测试模式）
在测试模式下进行气体测试、警报测试等各种测试。
项目编号
2-0
2-0.0
2-0.1
2-0.2
2-0.3
2-0.4

设定项目
测试模式
气体测试
警报测试
故障测试
LED 测试
RAM 测试

7-3-1 进行气体测试进行气体测试。进行气体测试之前，请准备气体测试用的气体，并连接到本仪器。
关于气体测试的准备，请参阅“7-4-1 气体校正的准备”。

警告• 气体测试即使在维护模式下，由于不同的设定，有时接点、气体浓度输出也会运行。操作时请注意。
1

在维护模式下多次按下▲键或▼键，选择[2-0]，
再按下 SET 键2选择［2-0.0］，按下 SET 键

3

导入气体测试用的气体4

气体浓度稳定后，停止气体测试用的气体

5指示值恢复[0.0]（氧气则为[20.9](vol%)）后，
按下 MENU/ESC 键

返回显示[2-0.0]。

111 / 171

7. 保养检查

7-3 进行各种测试（测试模式）

7-3-2 进行警报测试测试当指示值达到警报点时是否发出警报。

警告• 警报测试即使在维护模式下，由于不同的设定，有时接点、气体浓度输出也会运行。操作时请注意。

注记 本仪器的规格为无警报继电器时，也需要进行步骤 3 的操作。但是，即使选择[on]，接点也不会运行。

1

在维护模式下多次按下▲键或▼键，选择[2-0]，
再按下 SET 键

2按下▲键或▼键，选择[2-0.1]，再按下 SET 键

3

按下▲键或▼键，选择[oFF]或[on]，再按下 SET 键
使用的接点设定为警报、警报或故障时选择[on]，
除此以外选择[oFF]。

4按下▲键或▼键上下调整指示值

112 / 171

7. 保养检查

7-3 进行各种测试（测试模式）

警报灯(ALM1)

5

确认发出警报
指示值达到第一警报点，就发出第一警报。
发出第一警报后，警报灯(ALM1)点亮（红色）。

指示值达到第二警报点，就发出第二警报。
发出第二警报后，警报灯(ALM1)与警报灯(ALM2)点亮
（红色）
。

6

警报测试结束后，按下 MENU/ESC 键
返回显示[2-0.1]。

113 / 171

警报灯
(ALM1)

警报灯(ALM2)

7. 保养检查

7-3 进行各种测试（测试模式）

7-3-3 进行故障测试测试当本仪器发生故障时是否发出警报。

警告• 故障测试即使在维护模式下，由于不同的设定，有时接点、气体浓度输出也会运行。操作时请注意。

1

在维护模式下多次按下▲键或▼键，选择[2-0]，
再按下 SET 键

2按下▲键或▼键，选择[2-0.2]，再按下 SET 键

3

按下▲键或▼键，选择[on]，再按下 SET 键

4

确认发出故障警报发出故障警报后，故障灯(FAULT)点亮（黄色）。

5故障测试结束后，按下▲键或▼键，选择[oFF]，
再按下 SET 键
故障警报解除，故障灯(FAULT)熄灭。
按下 MENU/ESC 键也可以解除故障警报。

114 / 171

故障灯
(FAULT)7. 保养检查

7-3 进行各种测试（测试模式）

7-3-4 进行 LED 测试进行 LED 显示测试。
1

在维护模式下多次按下▲键或▼键，选择[2-0]，
再按下 SET 键

2按下▲键或▼键，选择[2-0.3]，再按下 SET 键

3

按下▲键或▼键，选择[on]，再按下 SET 键

4确认 LED 全部点亮

5LED 测试结束后，按下▲键或▼键，选择[oFF]，
再按下 SET 键
LED 测试结束。
按下 MENU/ESC 键也可以结束 LED 测试。

115 / 171

7. 保养检查

7-3 进行各种测试（测试模式）

7-3-5 进行存储器测试（RAM 测试）进行存储器测试。
1

在维护模式下多次按下▲键或▼键，选择[2-0]，
再按下 SET 键

2按下▲键或▼键，选择[2-0.4]，再按下 SET 键

3

按下 SET 键

存储器测试开始。

4确认存储器测试的结果存储器有异常时显示为[PASS]。

RAM 存储器有异常时显示为[F-rA]。

ROM 存储器有异常时显示为[F-rO]。

RAM 存储器和 ROM 存储器有异常时交替显示[F-rA]和[F-rO]。

5

确认结果后，按下 SET 键
返回显示[2-0.4]。

116 / 171

7. 保养检查

7-4 进行气体校正

7-4 进行气体校正7-4-1 气体校正的准备进行气体校正（零位校正、间距校正）之前，请准备校正用的气体。
此外，如下所示，请在连接本仪器和校正用的器材后开始校正。
 零位校正用气体 间距校正用气体(采集到气袋或气罐中) 排气用气体袋
 校正盖或防溅罩＜调整夹具列表＞
品名

校正盖

泵

流量计
秒表

适用规格
IRF 传感器用

部件编号：4283 9011 00

NCF/SGF/SHF 传感器用部件编号：4283 9012 70

ESF 传感器用

部件编号：4283 9013 40

流量达到 0.5mL 以上的防爆规格可测量容差 0.1ml
定期检查合格品
定期检查合格品
一般可燃性气体

配管

有机溶剂

高吸附性气体材质：聚氨酯
内径：φ4 ㎜
气体～校正盖之间 配管长度：1m 以内
材质：特氟龙
内径：φ4 ㎜
气体～校正盖之间 配管长度：1m 以内
材质：特氟龙
内径：φ4 ㎜
气体～校正盖之间 配管长度：10cm 以内

117 / 171

7. 保养检查

7-4 进行气体校正＜零位校正用气体＞
 零位校正用气体是校正零点的气体。周围空气中如果没有干扰杂质或测量对象气体，也可以用作零位校正气
体。O2 和 CO2 传感器使用氮气(N2)。

＜间距校正用气体＞
 间距校正用气体在间距校正中用于校正传感器的灵敏度。间距校正用气体是以清洁空气或氮气稀释的已知浓度
的气体。如为 O2 传感器，由于使用来自周围空气的氧气，因此不需要间距校正用气体。
 推荐使用检测对象气体校正仪器（实际气体校正）。这种方法比替代气体校正更准确。替代气体校正仅限在不能进行实际气体校正时方可作为替代手段执行。
 甲烷和氢气仅限使用实际气体进行校正，校正时请勿使用替代气体。
 间距校正自最后有效的零位校正起，需在 24 小时内执行。
 校正气体浓度的管理方法推荐使用本公司的便携检测仪。
 请联系经销商或就近的本公司营业网点。

警告• 间距校正用气体使用可燃性气体或毒性气体。一旦误吸入可能会损害健康或导致死亡，因此禁止吸入校正
气体。

118 / 171

7. 保养检查

7-4 进行气体校正＜使用校正盖时＞
注记 进行校正时，请将流量控制为 0.5 ± 0.1L/min。如果使用不同的流量进行校正，可能会降低校正精度。
 校正盖根据型号分成 3 种：IRF 传感器用、NCF/SHF/SGF 传感器用、ESF 用。
请注意，SD-3GHS/SD-3DGHS/GD-3GHS 作为例外，使用 IRF 传感器用的校正盖。
 如果装上了抽吸盖和校正盖，在无风状态下静置 10 分钟以上，吸附于配管或盖子上的气体可能会滞留，引起指示值上升（如果使用氧气，配管或盖子内变为缺氧，指示值降低）。此时请导入清洁空气，使指示值恢复为零。在此状态下传感器与大气被隔断，因此变为无监测状态。

＜使用气袋时＞

RI

EC

本仪器(SD-3)

流量计
泵
OUT

P

校正盖
针阀
气体袋

IN

119 / 171至排气

7. 保养检查

7-4 进行气体校正＜使用气袋时＞

NC

GH

SP

本仪器(SD-3)

校正盖
IN
流量计
泵
至排气

P

OUT
针阀

※ 图为 SD-3。遥控传感器头(GD-3)的连接方法也相同。

120 / 171气体袋

7. 保养检查

7-4 进行气体校正＜使用气罐时＞

RI

EC

本仪器(SD-3)

OUT
至排气

IN

＜使用气罐时＞

NC

GH

校正盖

SP

本仪器(SD-3)

IN

OUT

校正盖

至排气

※ 图为 SD-3。遥控传感器头(GD-3)的连接方法也相同。

121 / 171

7. 保养检查

7-4 进行气体校正＜使用防溅罩时＞

警告• 如果使用防溅罩，可能会影响气体响应时间。
• 气体的响应时间可能会变长，发出警报的时间变长。

注记 使用防溅罩时不需要校正盖。
 使用防溅罩时，请按照以下风速进行校正。
检测原理

风速

非分散型红外线吸收法（红外线式）
新型陶瓷式

6m/s 以下

半导体式热线型半导体式

1m/s 以下

恒电位电解式 如果校正失败，请替换为校正盖后进行校正。
 如果校正吸附性气体，请替换为校正盖后进行校正。
 进行校正时，请将流量控制为 0.5 ± 0.1L/min。如果使用不同的流量进行校正，可能会降低校正精度。

＜使用气袋时＞

本仪器(SD-3)

防溅罩

流量计
泵
气体袋

P
针阀

※ 图为 SD-3。遥控传感器头(GD-3)的连接方法也相同。

122 / 171

7. 保养检查

7-4 进行气体校正＜使用气罐时＞

防溅罩IN※ 图为 SD-3。遥控传感器头(GD-3)的连接方法也相同。

＜使用过滤器时＞

警告• 如果安装过滤器，可能会影响灵敏度、气体响应时间和替代气体灵敏度。
• 如果使用过滤器，气体的响应时间可能会变长，发出警报的时间变长。

＜使用保护罩时＞

警告• 如果安装保护罩，可能会影响灵敏度、气体响应时间和替代气体灵敏度。
• 如果使用保护罩，气体的响应时间可能会变长，发出警报的时间变长。

123 / 171

7. 保养检查

7-4 进行气体校正7-4-2 进行零位校正
如果不正确进行零位校正，就无法准确测量气体浓度。
为了准确测量，请务必进行零位校正。
进行零位校正之前，请准备零位校正用气体，并连接到本仪器。
关于零位校正的准备，请参阅“7-4-1 气体校正的准备”。

警告• 使用周围空气进行零位校正时，请先确认周围是清洁的大气。如果在有杂质气体的状态下进行零位校正，将无法准确校正，实际发生气体泄漏时很危险。

注意• 请供应零位校正用气体，待指示值稳定后再进行零位校正。

注记 请按照规定的顺序进行零位校正和间距校正。如果不按照正确的顺序进行，可能会影响传感器特性。
GHF 传感器：变为先间距校正、后零位校正。
GHF 传感器以外：先零位校正、后间距校正。

124 / 171

7. 保养检查

7-4 进行气体校正1

在维护模式下多次按下▲键或▼键，选择[2-1]，
再按下 SET 键

2

导入零位校正用的气体，按下 SET 键

零位校正开始。

3

确认零位校正的结果
如果零位校正成功
将会显示[PASS]，3 秒后自动返回显示[2-1]。

零位校正失败时
将会显示[FAIL]，按下 SET 键，即返回显示[2-1]。

注记 由于零点（氧气则为 N2）在零附近大幅变动等原因，零位校正失败时，显示[FAIL]。此时零位校正
未完成。

7-4-3 进行间距校正为了正确测量气体浓度，进行必要的间距校正。
进行间距校正之前，请准备间距校正用的气体，并连接到本仪器。
关于间距校正的准备，请参阅“7-4-1 气体校正的准备”。

注记 进行间距校正需要专用夹具。建议委托本公司业务人员购买。

125 / 171

7. 保养检查

7-4 进行气体校正＜为标准量程时＞

注意• 请供应间距校正用的气体（警报设定值的 1.6 倍浓度或全量程的 40%为宜），待指示值稳定后再进行间距
校正。
• 在缺氧警报规格下，请进行调整，以使得对清洁空气的值变为[20.9] (vol%)。
1

在维护模式下多次按下▲键或▼键，选择[2-2]，
再按下 SET 键

2导入间距校正用的气体，按下 SET 键
指示值低于全量程的 10%时无法按下 SET 键。

3按下▲键或▼键，调整指示值配合导入的气体浓度，
按下 SET 键

间距校正开始。

4

确认间距校正的结果间距校正成功时显示[PASS]后，显示校正浓度的上限值。

间距校正失败时
显示[FAIL]。
5

按下 SET 键
返回显示[2-2]。

126 / 171

7. 保养检查

7-4 进行气体校正注记 由于指示值与导入的气体浓度相比有大幅变动等原因，导致间距校正失败时，显示[FAIL]。此时间距校正
未完成。
 间距校正失败时，确认供给的间距校正用气体浓度是否正确，或者气体校正用的夹具有无泄漏等。如未正确进行间距校正，请再次进行间距校正。
 如果已正确进行间距校正或重新校正后也无法校正间距时，可能是气体传感器寿命已到。请断开电源，
通知本公司业务人员。

＜为双量程时＞ NC

注意• 请供应间距校正用的气体，待指示值稳定后再进行间距校正。
• 如果进行双量程的间距校正，请继续进行 L 量程和 H 量程的间距校正。如果中途停止了间距校正，请再次
从 L 量程开始重新进行。

1

在维护模式下多次按下▲键或▼键，选择[2-2]，
再按下 SET 键

2导入间距校正用的气体，按下 SET 键
指示值低于全量程的 10%时无法按下 SET 键。

3按下▲键或▼键，调整指示值配合导入的气体浓度，
按下 SET 键

L 量程的间距校正开始。

4确认 L 量程的间距校正的结果· L 量程的间距校正成功时
显示[PASS]和校正浓度的上限值后，自动进行 H 量程的
间距校正。
· L 量程的间距校正失败时显示[FAIL]，自动进行 H 量程的间距校正。

127 / 171

7. 保养检查

7-4 进行气体校正5

导入间距校正用的气体，按下 SET 键
指示值低于全量程的 10%时无法按下 SET 键。

6按下▲键或▼键，调整指示值配合导入的气体浓度，
按下 SET 键

H 量程的间距校正开始。

7

确认 H 量程的间距校正的结果· H 量程的间距校正成功时显示[PASS]后，显示校正浓度的
上限值。

· H 量程的间距校正失败时显示[FAIL]。

8

按下 SET 键
返回显示[2-2]。

注记 由于指示值与导入的气体浓度相比有大幅变动等原因，导致间距校正失败时，显示[FAIL]。此时间距校正
未完成。
 间距校正失败时，确认供给的间距校正用气体浓度是否正确，或者气体校正用的夹具有无泄漏等。如未正确进行间距校正，请再次进行间距校正。
 如果已正确进行间距校正或重新校正后也无法校正间距时，可能是气体传感器寿命已到。请断开电源，
通知本公司业务人员。

128 / 171

7. 保养检查

7-5 进行环境设定 1

7-5 进行环境设定 1
在环境设定 1 中，可以设定以下项目。
项目编号
2-4

设定项目

备注

环境设定 1
2-4.0

INHIBIT 设定

2-4.1

警报点设定

2-4.2

警报延迟设定客户方通常不使用

2-4.3

警报运行设定客户方通常不使用

2-4.4

警报点限制器设定客户方通常不使用

2-4.5

警报方式设定客户方通常不使用

2-4.6

故障运行设定客户方通常不使用

2-4.7

零抑制值设定客户方通常不使用

2-4.8

零抑制方式设定

客户方通常不使用

2-4.9

密码

客户方通常不使用

2-4.A

日期时间设定7-5-1 INHIBIT 设定
如果 INHIBIT 设定为 ON，即使在检测模式下显示气体浓度，也不会向外部发送警报。主要在本仪器的各种测试和检查时使用。
初始设定为 OFF。
1

在维护模式下按下▲键或▼键，选择[2-4]，
再按下 SET 键2选择[2-4.0]，按下 SET 键

3

按下▲键或▼键，选择[oFF]或[on]，再按下 SET 键

返回[2-4.0]的画面显示。

129 / 171

7. 保养检查

7-5 进行环境设定 1

注记 如果 INHIBIT 设定为 ON，进入检测模式时，在 LED 中交替显示[InHI.]和[0.0]（气体浓度值）。

7-5-2 警报点设定设定气体警报的警报点（第一警报点、第二警报点）。
请在以下范围内设定警报点。
警报点第一警报点([AL1])
第二警报点([AL2])警报点范围
全量程的 10% - 第二警报点(H-HH)
可燃性气体
第一警报点 - 全量程的 60%(H-HH)
可燃性气体以外 第一警报点 - 全量程(H-HH)

1

在维护模式下按下▲键或▼键，选择[2-4]，
再按下 SET 键

2按下▲键或▼键，选择[2-4.1]，再按下 SET 键

3

按下▲键或▼键，选择[AL1]或[AL2]，再按下 SET 键
设定第一警报点时选择[AL1]，设定第二警报点时选择[AL2]。

4

按下▲键或▼键设定警报点，再按下 SET 键

返回显示[AL1]或[AL2]。

注记 警报点设定值可以在用户模式的设定值显示中确认。
（参阅“5-7-1 确认警报设定值”）

130 / 171

7. 保养检查

7-5 进行环境设定 1

7-5-3 日期时间设定
设定内部时钟的日期时间（年、月、日、时、分）。
1

在维护模式下按下▲键或▼键，选择[2-4]，
再按下 SET 键

2按下▲键或▼键，选择[2-4.A]，再按下 SET 键

3

按下▲键或▼键，设定年（公历）的后 2 位，
再按下 SET 键选中的项目（年（公历）的后 2 位）闪烁显示。

4按下▲键或▼键设定月，再按下 SET 键选中的项目（月）闪烁显示。

5按下▲键或▼键设定日，再按下 SET 键选中的项目（日）闪烁显示。

6按下▲键或▼键设定时（24 小时显示），再按下 SET 键选中的项目（时）闪烁显示。

7按下▲键或▼键设定分，再按下 SET 键选中的项目（分）闪烁显示。

返回显示[2-4.A]。

131 / 171

7. 保养检查

7-6 进行环境设定 3

7-6 进行环境设定 3
在环境设定 3 中，可以设定以下项目。
项目编号
2-6

设定项目

备注

环境设定 32-6.0

使用接点设定客户方通常不使用

2-6.1

励磁/非励磁设定

客户方通常不使用

2-6.2

维护时的外部输出客户方通常不使用

2-6.3

外部输出调整(4 - 20mA)

客户方通常不使用

2-6.4

警报测试时外部输出客户方通常不使用

2-6.5

故障时外部输出设定客户方通常不使用

2-6.6

双量程外部输出设定NC

132 / 171

7. 保养检查

7-6-1 双量程外部输出设定

7-6 进行环境设定 3

NC

设定双量程的外部输出范围。
（参阅“6-4 外部输出运行的＜气体浓度和外部输出示例（为双量程时）＞”）
可以设定以下任意一种。
 4 - 20mA（L 量程）/22mA（H 量程）([L4-2])
 4 - 20mA（H 量程）([H4-2])
 4 - 16mA（L 量程）/16 - 20mA（H 量程）([4-16])
初始设定为 4 - 16mA（L 量程）/16 - 20mA（H 量程）([4-16])。

注记 双量程外部输出设定仅限在检测原理为新型陶瓷式时可以设定。
（参阅“1-2-2 各型号的检测原理和检测对
象气体”
）

1

在维护模式下按下▲键或▼键，选择[2-6]，
再按下 SET 键

2按下▲键或▼键，选择[2-6.6]，再按下 SET 键

3

按下▲键或▼键，选择[4-16]或[L4-2]、[H4-2]的任意一个，
再按下 SET 键

返回显示[2-6.6]。

133 / 171

7. 保养检查

7-7 部件的更换

7-7 部件的更换
7-7-1 传感器的更换更换传感器后，需由专业维修人员确认运行情况。为了机器稳定动作与安全，请委托专业的维修人员。请委托经销
商或就近的本公司营业网点。

＜传感器更换周期列表＞
传感器型号

气体种类

化学式

检测浓度

传感器更换周期

IRF

-

-

-

5年

NCF

-

-

-

3年

SHF

-

-

-

3年

SGF

-

-

-

3年

ESF-B242

氨

NH3

75ppm

2年

ESF-B245氯

CL2

1.5ppm1年

ESF-A24P

一氧化碳

CO

150ppm3年

ESF-A24RP

硫化氢

H2S

100ppm3年

ESF-A24RH

硫化氢

H2S

30ppm

3年

ESF-A24A

二氧化氮

NO2

15ppm

3年

ESF-A24E

氯化氢

HCL

15ppm

3年

ESF-X24P2

氧气O2

25vol%

3年

注意• 更换传感器时，请先关闭本仪器的供给电源后再进行更换。
• 传感器安装方法请参阅“4-8 安装传感器”。
• 传感器更换后，请进行零位校正和间距校正。
• 传感器无需断开主机电源，可以在危险分类区域内更换。
• 如果之前安装了相同类型（相同部件号）的传感器，会保留仪器固有的配置（检测气体、测量范围、警报
点、零抑制等）。如属不同类型，将会上传新传感器出厂时的默认设定，覆盖仪器固有的配置。利用传感
器自动识别功能进行识别。关于传感器自动识别功能，请参阅“6-5-2 传感器自动识别功能”。
• 利用传感器更换功能菜单的“传感器更换”，可以在运行中更换传感器，主机不会发出故障信号。

注记 更换传感器请委托本公司营业部。传感器更换后需使用标准气体进行气体校正。

134 / 171

7. 保养检查

7-7 部件的更换

7-7-2 定期更换部件的更换
本仪器的建议更换部件如下所示。请以推荐更换周期为标准进行更换。

＜推荐更换部件列表＞
名称传感器护板
SD-3RI /GHS 用传感器护板SD-3NC/GH/SP 用传感器护板SD-3EC/ECS/ECB 用
除硅过滤器（型号：SI-8）
活性碳过滤器（型号：CF-8304）

推荐检查周期推荐更换周期

数量
（个/台）

备注

3-8年
6 个月

1
1年
1年

商品编号：
4283 0030 20
商品编号：4283 0040 10

注记 以上更换周期是大致标准，可能因使用条件而异。另外，该周期不表示保修期。更换时期可能根据定期
检查的结果而改变。

135 / 171

8. 关于储存、迁移及报废

8-1 储存或长期不使用时的处理8关于储存、迁移及报废
8-1 储存或长期不使用时的处理
请在以下条件下储存本仪器。
 常温、常湿、避免直射阳光的场所 不产生气体、溶剂、蒸气等的场所＜储存条件（传感器）＞
型号

存储
温度
期限
(℃)
（年）
最小 最大

湿度※1
(%RH)

压力
(kPa)

最小

最大

最小

最大

SGF

0.5

-10

40

0

90

80

120

SHF

0.5

-10

40

0

90

80

120

NCF

0.5

-10

40

0

90

80

120

IRF

ESF

0.5

0.5

-10

-10

40

40

0

0

90

90

80

80

120

120

※1 湿度最大值为绝对湿度 33g/m3 以下(33.6℃ 90%RH)。

136 / 171备注
· 传感器应收入指定的包装箱内储存。
· 应储存在避开直射阳光的场所。
· 应储存在没有腐蚀性气体、振动、灰尘等的环境内。
· 应无结露或溅水。
· 储存环境中不应存在可能形成催化剂毒的气体（代表例：硅类、氯类、硫类）。

· 传感器应收入指定的包装箱内储存。
· 应储存在避开直射阳光的场所。
· 应储存在没有腐蚀性气体、振动、灰尘等的环境内。
· 应无结露或溅水。
· 传感器应收入指定的包装箱内储存。
· 应储存在避开直射阳光的场所。
· 应储存在没有腐蚀性气体、振动、灰尘等的环境内。
· 应无结露或溅水。
· 储存环境中不应存在可能形成催化剂毒的气体（代表例：硅类、氯类、硫类）。

8. 关于储存、迁移及报废

8-1 储存或长期不使用时的处理＜储存条件（传感器搭载 SD-3）＞
存储
期限
（年）
-

最小

最大

最小

最大

最小

最大

SGF

0.5

-10

40

0

90

80

120

SHF

0.5

-10

40

0

90

80

120

NCF

0.5

-10

40

0

90

80

120

型号

IRF

湿度※1
(%RH)

温度
(℃)

压力
(kPa)

0.5

-10

40

0

90

80

120

0.5

-10

40

0

90

80

120

0.5

-10

40

0

90

80

120

ESF-A※2

ESF-X※3

ESF-B※4备注· 应储存在避开直射阳光的场所。
· 应储存在没有腐蚀性气体、振动、灰尘等的环境内。
· 应无结露或溅水。
· 储存环境中不应存在可能形成催化剂
毒的气体（代表例：硅类、氯类、硫类）。
· 应储存在避开直射阳光的场所。
· 应储存在没有腐蚀性气体、振动、灰尘等的环境内。
· 应无结露或溅水。
· 应储存在避开直射阳光的场所。
· 应储存在没有腐蚀性气体、振动、灰尘等的环境内。
· 应无结露或溅水。
· 储存环境中不应存在可能形成催化剂
毒的气体（代表例：硅类、氯类、硫类）。
· 应储存在专用的传感器储存容器内※5。
使用时，在运转之前打开专用容器，取出传感器，将其安装到检测器。

※1 湿度最大值为绝对湿度 33g/m3 以下(33.6℃ 90%RH)。
※2 ESF-A 表示型号以“ESF-A”开头的传感器。
※3 ESF-X 表示型号以“ESF-X”开头的传感器。
※4 ESF-B 表示型号以“ESF-B”开头的传感器。
※5 如果不储存在容器内，在 90%RH 以上的环境中可能会发生漏液。此外，ESF-B※4 可能在低湿度下发生液体干涸。

注意• 本仪器很重并且有突起，一般的包装可能会损坏本仪器和同箱的构件。储存本仪器或由于任何原因将其退
回给本公司时，请使用装过本仪器的原包装。如果没有包装箱，请进行双层包装，并用硬物填满空隙，
以防止本仪器在包装箱内移动。此外，请勿装入易碎物品。

137 / 171

8. 关于储存、迁移及报废

8-2 移机或重新使用时的处理

8-2 移机或重新使用时的处理
本仪器移机并且重新使用时，关于搬迁场所，请参阅“4-1 有关安装的注意事项”。
此外，关于安装方法，请参阅“4. 安装方法”。

注意• 移机或停用储存本仪器后重新使用时，请务必进行气体校正。重新调试时（包括气体校正）请联系本公司
营业网点。

8-3 产品的废弃・・・・废弃本仪器时，请作为工业废弃物（不燃物）根据地方法律法规等进行妥善处理。
请务必将用过的传感器单元归还本公司。请退至本公司营业部。
恒电位电解式传感器单元（ESF 传感器）万一发生漏液时，请避免直接接触液体，务必将其放入塑料袋，避免
液体外漏。
本仪器和 ESF 传感器使用纽扣型电池。废弃时事先需进行妥善处理，请联系就近的代理商。

＜SD-3 主机＞
纽扣型锂电池的规格
品名

锂电池

公称电压

3.0V

公称容量

48mAh

拆卸方法
拆下右图红框所示的主电路板单元。拆下电路板支架，主电路板的背面（7 段数码管和反面）有电池。
电池以树脂外壳覆盖。树脂外壳粘附在电路板上，因此请用钳子或一字螺丝刀等工具拆下树脂外壳。
请使用钳子剪断两侧的端子，取出电池。请用带子缠绕电池处理取出的电池，以避免电池短路。

主电路板

138 / 171

树脂外壳

电路板支架

8. 关于储存、迁移及报废

8-3 产品的废弃＜ESF 传感器＞
纽扣型锂电池的规格
品名二氧化锰锂电池

公称电压

3.0V

公称容量

140mAh拆卸方法

ESF 传感器拆下 ESF 传感器的后盖（白色盖子）。
拆下的后盖上有电池，请将其取下。
请用带子缠绕处理取出的电池，以避免电池短路。

后盖

电池

警告
• ESF 传感器内含电解液，因此切勿拆解。如果接触了电解液，可能导致皮肤糜烂，进入眼睛则可能导致失明。此外，粘附在衣服上可能导致变色、破洞。万一接触了电解液时，请立刻用水充分清洗接触的部分。

注记 本仪器内置电池。
 关于封闭式回收垃圾箱标志
本标志标注在执行欧盟电池指令 2006/66/EC 的内置电池的产品上，
需按照妥善的方法废弃电池。
本标志表示废弃电池时，需要与一般垃圾分类处理。

139 / 171

9. 故障排除9-1 仪器异常9

故障排除
本故障排除章节并未罗列全部故障。本章节简单描述了有助于查明常见故障原因的内容。
如果您对本章节所述的故障表现采取相应措施但仍未能恢复，请联系经销商或就近的本公司销售网点。

9-1 仪器异常状况、显示

故障灯(FAULT)无法接通电源

原因

对策

配线错误

请正确配线。

配线从端子排脱落请正确连接。

电源系统的异常、瞬断请确认是否供给了额定电压。
请采取修改或增加无停电电源、电源线路滤波器、绝缘变压器等措施。

线缆的异常（断线、未连接、短路）

请确认包括本仪器及周边相关仪器在内的配线
情况。

保险丝断线

需更换。请联系本公司营业网点。

内部电路异常需更换。请联系本公司营业网点。

突发的噪声等的影响请暂时先停止电源供给，重新启动。频繁发生这种状况时，请正确实施防噪对策。
如果实施了噪声对策仍未见改善，请联系本公司营业网点。

传感器未连接或者连接不良请停止电源供给，确认传感器的连接状态。

传感器发生了故障

请更换新的传感器。

突发的电涌噪声等的影响请暂时先停止电源供给，重新启动。频繁发生这种状况时，请正确实施防噪对策。
如果实施了噪声对策仍未见改善，请联系本公司营业网点。

（点亮） 主机(SD-3、SD-3SC)或遥控传感器头(GD-3)的内部配线错误请正确完成主机(SD-3、SD-3SC)或遥控传感器头(GD-3)的内部配线。

-

运行异常
-

传感器异常
（E-1）

主机(SD-3SC)和遥控传
感器头(GD-3)之间的线缆异常

140 / 171

请确认主机(SD-3SC)和遥控传感器头(GD-3)之间的线缆。

9. 故障排除状况、显示

9-2 指示值异常故障灯(FAULT)

传感器异常
(E-1)

（点亮）

系统异常
(E-9)

原因

对策

主机(SD-3、SD-3SC)或遥控传感器头(GD-3)的保险丝断线需更换。请联系本公司营业网点。

主机(SD-3、SD-3SC)或遥控传感器头(GD-3)的内部电路异常需更换。请联系本公司营业网点。

4 - 20mA 输出断线请停止电源供给，确认 4 - 20mA 输出的连接状态。

仪器内部的 ROM、（点亮） RAM、EEPROM 故障
GD-3 异常
(E-1A)

请暂时先停止电源供给，重新启动。如此处理后仍没有恢复时，需进行更换。请联系本公司
营业网点。

传感器未连接或者连接不良请停止电源供给，确认传感器的连接状态。

传感器发生了故障

请更换新的传感器。

突发的电涌噪声等的影响请暂时先停止电源供给，重新启动。频繁发生这种状况时，请正确实施防噪对策。
如果实施了噪声对策仍未见改善，请联系本公司营业网点。

原因

对策

（点亮）

9-2 指示值异常
状况、显示

故障灯
(FAULT)·指示值上升（下降）后不恢复
·检测场所无气体
泄漏等异常却发出气体警报
·响应迟缓-传感器的零位或间距发生了变化（漂移）

请进行零位校正或间距校正。

存在干扰气体请确认有无溶剂等的干扰气体，进行妥善
处理。

缓慢泄漏

可能是待检测气体有微量泄漏。如果放置不管，可能发生危险的状态，因此请按照气体警
报时的措施，采取同等措施进行处理。

环境变化

请进行零位校正或间距校正。

噪声的影响

请暂时先停止电源供给，重新启动。频繁发生这种状况时，请正确实施防噪对策。
如果实施了噪声对策仍未见改善，请联系本公司营业网点。

环境骤变

本仪器可能因急剧的环境变化（温度等）而出
现警报指示。
如果环境频繁急剧变化，本仪器就无法使用。
请客户采取妥善措施。

传感器灵敏度劣化

请更换新的传感器。

校正气体浓度不正确请准备正确的校正气体。

传感器灵敏度劣化

请更换新的传感器。

141 / 171

9. 故障排除状况、显示

9-2 指示值异常故障灯(FAULT)

原因

对策

时钟异常

请进行日期时间设定。
使用 EC 的灵敏度校正功能时可能无法正常实施
校正。
如果经常出现这种状况，可能是内部时钟出现
故障。请换上新的内部时钟。
请联系本公司营业网点。

-

本仪器安装场所的环境
温度脱离了使用温度
范围本仪器的使用温度范围因工作原理而异。
如果环境温度脱离使用温度范围，本仪器无法使用。
请客户采取妥善措施。

传感器使用寿命
警告
(E-8)

-

传感器使用寿命即将结束建议更换传感器。
请联系本公司营业网点。

传感器警告
(E-1)

-

系统异常
(E-9)
-

超出使用温度范围(E-27)

请联系本公司营业网点。

142 / 17110. 产品规格

10-1 SD-3 规格

10

产品规格
10-1 SD-3 规格
10-1-1 SD-3（无 EC 安全栅规格）
型号
规格
显示
传感器
检测原理
检测对象气体
检测量程
采样方式
设定流量
电源显示
外部输出
警报设定值
警报精度
警报延迟时间
重复性（相同条件下）指示精度（同一条件下）气体警报类型气体警报显示气体警报复位运行
故障警报、自我诊断故障警报显示故障警报复位运行

接点动作
警告
警告显示
警告运行
各种功能

SD-3
标准

SD-3（接点规格）
触点

7 段数码管 LED(5 位)F 传感器（理研计器制）红外线式、恒电位电解式、新型陶瓷式、半导体式、热线型半导体式取决于传感器规格取决于传感器规格扩散式/抽吸式（通过外部单元导入）0.4～1.5L/min
电源指示灯点亮（绿）气体浓度信号（DC4 - 20mA with HART 输出）
取决于传感器规格取决于传感器规格取决于传感器规格取决于传感器规格取决于传感器规格
2 级警报（H-HH 或 H-L、L-LL）警报灯点亮（红）自动复位或自我保持系统异常(E-9)/传感器异常(E-1)
故障灯点亮（黄）/错误编号显示系统异常：自我保持传感器异常：自动复位（但是，无传感器时为自我保持）
SPDT(× 3)：
2 警报、1 故障输出运行―
AC 250V 2A、DC 30V 1A（电阻负载）
最小负载 DC 5V 0.1 A
传感器寿命诊断/时钟异常诊断/通信诊断/传感器警告气体浓度值和错误编号交替闪烁显示
与常规运行相同警报延迟/抑制/HART 通信

143 / 17110. 产品规格

型号
传输方式(4 - 20mA)
传输规格(4 - 20mA)
传输线缆(4 - 20mA)
传输距离(4 - 20mA)
电源

电源线缆
功耗
线缆连接口
操作方法
使用温度范围
使用湿度范围
外壳材质
保护等级
安装方法
防爆结构
防爆等级
外形尺寸
（突起部分除外）重量10-1 SD-3 规格SD-3
SD-3（接点规格）
3 线式模拟传输（电源通用<电源、信号、公共端>）或2 线式模拟传输（电流源）DC 4 - 20mA（非绝缘/线性输出）
负载电阻 600Ω以下（有基于供给电压的降额）分辨率最大 250 分割（取决于规格）
带屏蔽的线缆 1.25sq (1.308 mm2/AWG16)或2.0sq (2.08 mm2/AWG14)（与电源线通用）※ 考虑到防爆，与使用环境中的预期最高温度相比，请使用耐热性至少高 5℃的线缆。
为 1.25sq (1.308mm2/AWG16)时：1.25km 以下
为 2.0sq (2.08mm2/AWG14)时：2 km 以下（有基于供给电压的降额）DC 24V（DC 18V - DC 30V）
※ 为了在仪器异常时正常断开保险丝，请使用可承受 2.5A 以上瞬时输出的电源。
带屏蔽的线缆 1.25sq (1.308mm2/AWG16)或2.0sq (2.08mm2/AWG14)（与传输线缆通用）※ 考虑到防爆，与使用环境中的预期最高温度相比，请使用耐热性至少高 5℃的线缆。
参阅“10-1-3 电气额定值(SD-3)”
M25 × 1.5专用电磁式控制钥匙
-40℃ - +70℃（无骤变）如果基于传感器规格存在限制，应遵守传感器规格
0%RH - 95%RH 以下（无结露）如果基于传感器规格存在限制，应遵守传感器规格不锈钢(SCS14)相当于 IP66/67
壁挂（标准）/2B 杆安装（选配件）耐压防爆结构
取决于每种型号
参阅“2-4 安全信息”的＜防爆性能数据＞
扩散式：约 171(W) × 277(H) × 127(D) mm
抽吸式：约 171(W) × 289(H) × 127(D) mm
扩散式：约 6.7 kg/抽吸式：约 7.0kg※ 请注意如果在接点闭合状态下使用，振动等原因可能导致接点粘连。

144 / 17110. 产品规格

10-1 SD-3 规格＜尺寸图（SD-3（无 EC 安全栅规格）
）＞
＜扩散式＞

＜抽吸式＞

145 / 17110. 产品规格

10-1 SD-3 规格

10-1-2 SD-3（有 EC 安全栅规格）
型号

SD-3

SD-3（接点规格）

规格
显示
传感器
检测原理
检测对象气体
检测量程
采样方式
设定流量
电源显示
外部输出
警报设定值
警报精度
警报延迟时间
重复性（相同条件下）指示精度（同一条件下）气体警报类型气体警报显示气体警报复位运行
故障警报、自我诊
断故障警报显示标准

触点

故障警报复位运行

接点动作
警告
警告显示
警告运行
各种功能
传输方式
(4 - 20mA)
传输规格（4 - 20mA）
传输线缆(4 - 20mA)
传输距离(4 - 20mA)电源

7 段数码管 LED（5 位）F 传感器（理研计器制）恒电位电解式取决于传感器规格取决于传感器规格扩散式/抽吸式（通过外部单元导入）0.4～1.5L/min
电源指示灯点亮（绿）气体浓度信号（DC4 - 20mA with HART 输出）
取决于传感器规格取决于传感器规格取决于传感器规格取决于传感器规格取决于传感器规格
2 级警报（H-HH 或 H-L、L-LL）警报灯点亮（红）自动复位或自我保持系统异常(E-9)/传感器异常(E-1)
故障灯点亮（黄）/错误编号显示系统异常：自我保持传感器异常：自动复位（但是，无传感器时为自我保持）
SPDT(× 3)：
2 警报、1 故障输出运行―
AC 250V 2A、DC30V 1A（电阻负载）
最小负载 DC 5V 0.1A
传感器寿命诊断/时钟异常诊断/通信诊断/传感器警告气体浓度值和错误编号交替闪烁显示
与常规运行相同警报延迟/抑制/HART 通信
3 线式模拟传输（电源通用<电源、信号、公共端>）或2 线式模拟传输（电流源）DC 4 - 20mA（非绝缘/线性输出）
负载电阻 600Ω以下（有基于供给电压的降额）分辨率最大 250 分割（取决于规格）
带屏蔽的线缆 1.25sq (1.308mm2/AWG16)或2.0sq (2.08mm2/AWG14)（与电源线通用）※ 考虑到防爆，与使用环境中的预期最高温度相比，请使用耐热性至少高 5℃的线缆。
为 1.25sq (1.308mm2/AWG16)时：1.25km 以下
为 2.0sq (2.08mm2/AWG14)时：2 km 以下（有基于供给电压的降额）DC 24V (DC 18V - DC 30V)
※ 为了在仪器异常时正常断开保险丝，请使用可承受 2.5A 以上瞬时输出的电源。

146 / 17110. 产品规格

型号

10-1 SD-3 规格SD-3

SD-3（接点规格）
2

电源线缆
功耗
线缆连接口
操作方法
使用温度范围
使用湿度范围
外壳材质
保护等级
安装方法
防爆结构
防爆等级
外形尺寸
（突起部分除外）重量带屏蔽的线缆 1.25sq (1.308 mm /AWG16)或2.0sq (2.08 mm2/AWG14)（与传输线缆通用）※ 考虑到防爆，与使用环境中的预期最高温度相比，请使用耐热性至少高 5℃的线缆。
参阅“10-1-3 电气额定值(SD-3)”
M25 × 1.5专用电磁式控制钥匙
-40℃ - +70℃（无骤变）如果基于传感器规格存在限制，应遵守传感器规格
0%RH - 95%RH 以下（无结露）如果基于传感器规格存在限制，应遵守传感器规格不锈钢(SCS14)相当于 IP66/67
壁挂（标准）/2B 杆安装（选配件）耐压防爆结构+本质安全防爆结构取决于每种型号
参阅“2-4 安全信息”的＜防爆性能数据＞
扩散式：约 171(W) × 322(H) × 127(D) mm
抽吸式：约 171(W) × 334(H) × 127(D) mm
扩散式：约 7.3kg；抽吸式：约 7.6kg※ 请注意如果在接点闭合状态下使用，振动等原因可能导致接点粘连。

147 / 17110. 产品规格

10-1 SD-3 规格＜尺寸图（SD-3（有 EC 安全栅规格）
）＞
＜扩散式＞

＜抽吸式＞

148 / 17110. 产品规格

10-1 SD-3 规格

10-1-3 电气额定值(SD-3)
型号

供给电源DC 24V (DC 18V - DC 30V)

最高 3.8W

SD-3GHDC 24V (DC 18V - DC 30V)

最高 4.5W

SD-3GHSDC 24V (DC 18V - DC 30V)

最高 4.5W

SD-3NCDC 24V (DC 18V - DC 30V)

最高 4.5W

SD-3SP

DC 24V (DC 18V - DC 30V)

最高 3.5W

SD-3ECDC 24V (DC 18V - DC 30V)

最高 2.8W

SD-3ECS

DC 24V (DC 18V - DC 30V)

最高 2.8W

SD-3ECBDC 24V (DC 18V - DC 30V)

最高 3.1W

SD-3RI

149 / 171功耗

10. 产品规格

10-2 SD-3SC＋GD-3 规格

10-2 SD-3SC＋GD-3 规格
10-2-1 SD-3SC＋GD-3（无 EC 安全栅规格）型号
规格SD-3SC+GD-3

SD-3SC（接点规格）+GD-3

标准

触点

显示

7 段数码管 LED（5 位）传感器F 传感器（理研计器制）检测原理红外线式、恒电位电解式、新型陶瓷式、半导体式、热线型半导体式检测对象气体取决于传感器规格检测量程取决于传感器规格

采样方式

扩散式

电源显示

电源指示灯点亮（绿）外部输出气体浓度信号（DC4 - 20mA with HART 输出）

警报设定值取决于传感器规格警报精度

取决于传感器规格警报延迟时间

取决于传感器规格重复性（相同条件下）取决于传感器规格

指示精度（相同条件下）取决于传感器规格气体警报类型

2 级警报（H-HH 或 H-L 或 L-LL）气体警报显示警报灯点亮（红）气体警报复位运行自动复位或自我保持

故障警报、自我诊断

系统异常(E-9)/传感器异常(E-1)

故障警报显示

故障灯点亮（黄）/错误编号显示

故障警报复位运行系统异常：自我保持传感器异常：自动复位（但是，无传感器时为自我保持）
SPDT(× 3)：
2 警报、1 故障输出运行
AC 250V 2A、DC 30V 1A（电阻负载）
最小负载 DC 5V 0.1A

接点动作

―

警告

传感器寿命诊断/时钟异常诊断/通信诊断/传感器警告警告显示

气体浓度值和错误编号交替闪烁显示

警告运行与常规运行相同各种功能警报延迟/抑制/HART 通信

传输方式(4 - 20mA)

3 线式模拟传输（电源通用<电源、信号、公共端>）或、
2 线式模拟传输（电流源）传输规格(4 - 20mA)

DC 4 - 20mA（非绝缘/线性输出）
负载电阻 600Ω以下（有基于供给电压的降额）分辨率最大 250 分割（取决于规格）

150 / 17110. 产品规格型号
传输线缆
(4 - 20mA)

传输距离(4 - 20mA)
传输线缆
（GD-3 之间）

10-2 SD-3SC＋GD-3 规格

SD-3SC+GD-3

SD-3SC（接点规格）+GD-3

带屏蔽的线缆 1.25sq (1.308mm2/AWG16)或2.0sq (2.08mm2/AWG14)（与电源线通用）※ 考虑到防爆，与使用环境中的预期最高温度相比，请使用耐热性至少高 5℃的线缆。
为 1.25sq (1.308mm2/AWG16)时：1.25km 以下
为 2.0sq (2.08mm2/AWG14)时：2km 以下（有基于供给电压的降额）带屏蔽双绞线 1.25sq (1.308mm2/AWG16)※ 考虑到防爆，与使用环境中的预期最高温度相比，请使用耐热性至少高 5℃的
线缆。

传输距离
（GD-3 之间）

20m (MAX)电源DC 24V（DC 18V - DC 30V）
※ 为了在仪器异常时正常断开保险丝，请使用可承受 2.5 A 以上瞬时输出的电源。

电源线缆带屏蔽的线缆 1.25sq (1.308mm2/AWG16)或2.0sq (2.08 mm2/AWG14)（与传输线缆通用）※ 考虑到防爆，与使用环境中的预期最高温度相比，请使用耐热性至少高 5℃的线缆。

功耗参阅“10-2-3 电气额定值(SD-3SC+GD-3)”

线缆连接口

M25 × 1.5

操作方法

专用电磁式控制钥匙

使用温度范围

-40℃ - +70℃（无骤变）如果基于传感器规格存在限制，应遵守其规格

使用湿度范围

0%RH - 95%RH 以下（无结露）如果基于传感器规格存在限制，应遵守其规格外壳材质

不锈钢(SCS14)保护等级

相当于 IP66/67

安装方法

壁挂（标准）/2B 杆安装（选配件）防爆结构耐压防爆结构

防爆等级取决于每种型号
参阅“2-4 安全信息”的＜防爆性能数据＞

外形尺寸(SD-3SC)

约 171(W) × 193(H) × 127(D) mm（突起部分除外）

重量(SD-3SC)

约 6.0kg

外形尺寸(GD-3)

约 125(W) × 195(H) × 88(D) mm
约 3.0kg

重量(GD-3)※ 请注意如果在接点闭合状态下使用，振动等原因可能导致接点粘连。

151 / 17110. 产品规格

10-2 SD-3SC＋GD-3 规格＜尺寸图（SD-3SC＋GD-3（无 EC 安全栅规格）
）＞

＜扩散式＞

152 / 17110. 产品规格

10-2 SD-3SC＋GD-3 规格

10-2-2 SD-3SC＋GD-3（有 EC 安全栅规格）型号
规格SD-3SC+GD-3

SD-3SC（接点规格）+GD-3

标准

触点

显示

7 段数码管 LED（5 位）传感器F 传感器（理研计器制）检测原理

恒电位电解式检测对象气体取决于传感器规格检测量程取决于传感器规格

采样方式

扩散式

电源显示

电源指示灯点亮（绿）外部输出气体浓度信号（DC4 - 20mA with HART 输出）

警报设定值取决于传感器规格警报精度

取决于传感器规格警报延迟时间

取决于传感器规格重复性（相同条件下）取决于传感器规格

指示精度（同一条件下）取决于传感器规格气体警报类型

2 级警报（H-HH 或 H-L 或 L-LL）气体警报显示电源指示灯点亮（红）气体警报复位运行自动复位或自我保持

故障警报、自我诊断

系统异常(E-9)/传感器异常(E-1)

故障警报显示

故障灯点亮（黄）/错误编号显示

故障警报复位运行系统异常：自我保持传感器异常：自动复位（但是，无传感器时为自我保持）
SPDT(× 3)：
2 警报、1 故障输出运行
AC 250V 2A、DC 30V 1A（电阻负载）
最小负载 DC 5V 0.1A

接点动作

―

警告

传感器寿命诊断/时钟异常诊断/通信诊断/传感器警告警告显示

气体浓度值和错误编号交替闪烁显示

警告运行与常规运行相同各种功能警报延迟/抑制/HART 通信

传输方式(4 - 20mA)

3 线式模拟传输（电源通用<电源、信号、公共端>）或2 线式模拟传输（电流源）传输规格(4 - 20mA)

DC 4 - 20mA（非绝缘/线性输出）
负载电阻 600Ω以下（有基于供给电压的降额）分辨率最大 250 分割（取决于规格）

传输线缆(4 - 20mA)

带屏蔽的线缆 1.25sq (1.308mm2/AWG16)或2.0sq (2.08mm2/AWG14)（与电源线通用）※ 考虑到防爆，与使用环境中的预期最高温度相比，请使用耐热性至少高 5℃的线缆。

153 / 17110. 产品规格

型号
传输距离(4 - 20mA)
传输线缆
（GD-3 之间）

10-2 SD-3SC＋GD-3 规格

SD-3SC+GD-3

SD-3SC（接点规格）+GD-3

为 1.25sq (1.308mm2/AWG16)时：1.25 km 以下
为 2.0sq (2.08mm2/AWG14)时：2km 以下（有基于供给电压的降额）带屏蔽双绞线 1.25sq (1.308mm2/AWG16)※ 考虑到防爆，与使用环境中的预期最高温度相比，请使用耐热性至少高 5℃的线缆。
20m (MAX)

传输距离（GD-3 之间）电源DC 24V (DC 18V - DC 30V)
※ 为了在仪器异常时正常断开保险丝，请使用可承受 2.5A 以上瞬时输出的电源。

电源线缆带屏蔽的线缆 1.25sq (1.308mm2/AWG16)或2.0sq (2.08mm2/AWG14)（与传输线缆通用）※ 考虑到防爆，与使用环境中的预期最高温度相比，请使用耐热性至少高 5℃的线缆。

功耗参阅“10-2-3 电气额定值(SD-3SC+GD-3)”

线缆连接口

M25 × 1.5

操作方法

专用电磁式控制钥匙

使用温度范围

-40℃ - +70℃（无骤变）如果基于传感器规格存在限制，应遵守传感器规格

使用湿度范围

0%RH - 95%RH 以下（无结露）如果基于传感器规格存在限制，应遵守传感器规格外壳材质

不锈钢(SCS14)保护等级

相当于 IP66/67

安装方法

壁挂（标准）/2B 杆安装（选配件）防爆结构耐压防爆结构+本质安全防爆结构防爆等级取决于每种型号
参阅“2-4 安全信息”的＜防爆性能数据＞

外形尺寸(SD-3SC)

约 171(W) × 193(H) × 127(D) mm（突起部分除外）

重量(SD-3SC)

约 6.0 kg

外形尺寸(GD-3)

约 125(W) × 240(H) × 88(D) mm
约 4.0kg

重量(GD-3)※ 请注意如果在接点闭合状态下使用，振动等原因可能导致接点粘连。

154 / 17110. 产品规格

10-2 SD-3SC＋GD-3 规格＜尺寸图（SD-3SC＋GD-3（有 EC 安全栅规格）
）＞

＜扩散式＞

155 / 17110. 产品规格

10-2 SD-3SC＋GD-3 规格

10-2-3 电气额定值(SD-3SC+GD-3)
型号

供给电源

功耗

SD-3SCDC 24V (DC 18V - DC 30V)

最高 5W

GD-3RIDC 24V (DC 18V - DC 30V)

最高 1.2W

GD-3GHDC 24V (DC 18V - DC 30V)

最高 2W

GD-3GHSDC 24V (DC 18V - DC 30V)

最高 2W

GD-3NCDC 24V (DC 18V - DC 30V)

最高 2W

GD-3SP

DC 24V (DC 18V - DC 30V)

最高 1W

GD-3ECDC 24V (DC 18V - DC 30V)

最高 1W

GD-3ECS

DC 24V (DC 18V - DC 30V)

最高 1W

GD-3ECBDC 24V (DC 18V - DC 30V)

最高 1W

156 / 17110. 产品规格

10-3 各种原理的检测对象气体

10-3 各种原理的检测对象气体
10-3-1 非分散型红外线吸收法（红外线式）
传感器型号

气体名称

气体
符号

F.S.

1 数位

第一

第二

校正用气体

警报点

警报点

（替代气体）

IRF-1301※

甲烷

CH4

100%LEL

0.5%LEL

25%LEL

50%LELCH4

IRF-1303※

异丁烷

i-C4H10

100%LEL

0.5%LEL

25%LEL

50%LELi-C4H10

IRF-1317※

丙烷

C3H8

100%LEL

0.5%LEL

25%LEL

50%LELIRF-1435

二氧化碳

CO2

2000ppm

1ppm

1000ppm-

CO2

IRF-1436

二氧化碳

CO2

5000ppm

10ppm

1000ppm-

CO2

IRF-1433

二氧化碳

CO2

10000ppm

10ppm

1000ppm-

CO2

IRF-1437

二氧化碳

CO2

2vol%

0.005vol%

1vol%

-

CO2

使用
温度范围

使用
湿度范围

保修年数

C3H8
(i-C4H10)

IRF-1438

二氧化碳

CO2

5vol%

0.010vol%

1vol%

-

CO2

IRF-1439

二氧化碳

CO2

10vol%

0.01vol%

1vol%

-

CO2

IRF-1334※

甲烷

CH4100vol%0.5vol%25vol%

50vol%

CH4

IRF-1316※

乙烯

C2H4

100%LEL

0.5%LEL

25%LEL

50%LELIRF-1340※

异丁烯

i-C4H8

100%LEL

0.5%LEL

25%LEL

50%LELIRF-1308※

正己烷

n-C6H14

100%LEL

0.5%LEL

25%LEL

50%LELIRF-1332※

丁二烯

C4H6

100%LEL

0.5%LEL

25%LEL

50%LEL-40 70℃

95%RH
以下
（应无
结露）

3年

C2H4
(CH4)
i-C4H8(i-C4H10)
n-C6H14(i-C4H10)
C4H6
(CH4)

※ 存在其他碳化氢气体等的干扰。

注记 请在使用温度范围内，安装在没有急剧变化的稳定场所。
 进行校正时，请先暖机 10 分钟以上，再进行零位校正和间距校正。
 校正环境和测量环境的温度、湿度、压力请保持相同。
（指示值因温度特性、湿度特性、压力特性而改
变，某些情况下可能因传感器的特性而超过零抑制值。
）
 气体灵敏度较低的乙烯或丁二烯，建议将警报设定为 25%LEL 以上后使用。
 如果基底气体的成分在校正时和测量时差异很大，例如校正时基于 Air，测量时基于 CO2，由于红外线吸收特性，即使对象气体浓度相同，指示值也会有偏差。
 如果在使用寿命诊断、自我诊断功能时显示警告，请实施维护。
 实施校正时，请使用 IRF 传感器校正盖。如果不使用 IRF 传感器校正盖，校正精度可能会降低。
 NCF 传感器的保修年数与检测对象气体无关，一律 3 年。

157 / 17110. 产品规格

10-3 各种原理的检测对象气体

10-3-2 新型陶瓷式传感器型号

NCF-6318
NCF-6319

气体名称

甲烷
VCM

气体
符号
CH4
VCM

F.S.

1 数位

100%LEL

0.5%LEL

100%LEL

0.5%LEL

第一

第二

校正用

使用

使用湿度

警报点

警报点

气体

温度范围

范围

25%LEL

50%LEL

CH4※2

25%LEL

50%LELi-C4H10

保修年数

0 - 95%RH-40 - 70℃

以下※1

3年

(无结露)
NCF-6320

氢

H2

100%LEL

0.5%LEL

25%LEL

50%LELH2※3※1 绝对湿度 46g/m3 以下（等同 37℃ 95%RH）※2 如果要求进行甲烷校正，或检测对象气体为甲烷、丙烷、乙烷，请使用 NCF-6318。NCF-6318、NCF-6319 的检测对象气体重复，如果不属于上述情况，请使用 NCF-6319。
※3 如果要求氢气选择性，请使用 NCF-6320。

注记 请在使用温度范围内，安装在没有急剧变化的稳定场所。
 硅气体（有机硅类气体）或各种卤素气体、硫化物气体、酸性气体的存在可能会导致传感器特性发生显著劣化。
 可选配除硅过滤器和活性碳过滤器。
 规格为 F.S.5000ppm 以下时，请先在清洁的环境下暖机 120 分钟以上，再进行零位校正和间距校正。
 规格为 F.S.5000ppm 以下时，环境中温湿度变化的影响增大，零点可能会漂移。此时请将零跟踪功能设为 ON。但是，如果始终存在气体，则无法使用零跟踪功能，因此建议使用其他检测原理的型号。
 即使是检测对象气体以外的可燃性气体，指示值也可能发生变动。
 如果氧气浓度降低，指示值可能会降低。
 作为基底气体，如果同时存在高浓度的 CO2、Ar、He 等热传导率大的气体，可能会受到干扰的影响。
 如果使用抽吸法检测蒸气压低的气体，从高温侧吸入气体，在低温侧进行检测时，气体可能会雾化，导致传感器内部结露或因烧结而堵塞。使用时请注意蒸气压。
 如果在使用寿命诊断、自我诊断功能时显示警告，请实施维护。
 实施校正时，请使用可燃性传感器校正盖。如果不使用可燃性传感器校正盖，校正精度可能会降低。
 如果与进行零位校正或间距校正的环境温度相比，温度变化达到±40℃以上，则指示精度可能会降低。
此时请再次进行零位校正或间距校正。
 如果受到外部的强力冲击或振动，指示值可能会改变。此时请进行零位校正和间距校正。
 如果长时间持续接触 100%LEL 以上的高浓度气体，零点可能会改变。此时请进行零位校正和间距校正。
 更换传感器时，传感器可能会处于高温状态。更换传感器时请小心避免烫伤。如果传感器很烫，请断开电
源，静置直至温度降下来。
 传感器上面配有特氟龙膜。如果特氟龙膜损坏，可能会影响特性。请勿强力按压或以锐利的物体弄伤特氟龙膜。
 NCF 传感器的保修年数与检测对象气体无关，一律 3 年。

158 / 17110. 产品规格

10-3 各种原理的检测对象气体

10-3-3 半导体式
传感器型号

气体名称

气体
符号

F.S.

1 数位

第一

第二

校正用气

使用

使用

警报点

警报点

体

温度范围

湿度范围

SGF-8562

二硫化碳

CS2

200ppm

1ppm

50ppm

100ppmCS2

SGF-8563

氧化乙烯

EO

100ppm

1ppm

25ppm

50ppm

EO

SGF-8562

硫化氢

H2S

100ppm

1ppm

25ppm

50ppm

H2S

保修年数

20 - 95%RH-20 - 65℃

以下※

3年

（无结露）

※ 绝对湿度 46g/m3 以下（等同 37℃ 95%RH）

注记 硅气体（有机硅类气体）或各种卤素气体、硫化物气体、酸性气体的存在可能会导致传感器特性发生显著劣化。
 可选配除硅过滤器和活性碳过滤器。
 请根据使用环境的湿度制作校正用气体。
 进行校正时，请先在清洁的环境下根据无通电时间进行暖机，再进行零位校正和间距校正。暖机时间请参阅以下的＜半导体式传感器的暖机时间＞。
 即使是检测对象气体以外的可燃性气体，指示值也可能发生变动。
 指示值也可能因安装场所的环境变化（温度、湿度等）而发生变动。
 如果氧气浓度降低，指示值可能会降低。
 作为基底气体，如果同时存在高浓度的 CO2、Ar、He 等热传导率大的气体，可能会受到干扰的影响。
 如果使用抽吸法检测蒸气压低的气体，从高温侧吸入气体，在低温侧进行检测时，气体可能会雾化，导致传感器内部结露或因烧结而堵塞。使用时请注意蒸气压。
 如果在使用寿命诊断、自我诊断功能时显示警告，请实施维护。
 实施校正时，请使用可燃性传感器校正盖。如果不使用可燃性传感器校正盖，校正精度可能会降低。
 如果与进行零位校正或间距校正的环境温度相比，温度变化达到 ±20℃以上，则指示精度可能会降低。
此时请再次进行零位校正或间距校正。
 如果受到外部的强力冲击或振动，指示值可能会改变。此时请进行零位校正和间距校正。
 如果长时间持续接触高浓度气体，零点可能会改变。此时请进行零位校正和间距校正。
 在气体连续存在的环境下，指示值可能会降低。
 更换传感器时，传感器可能会处于高温状态。更换传感器时请小心避免烫伤。如果传感器很烫，请断开电
源，静置直至温度降下来。
 传感器上面配有特氟龙膜。如果特氟龙膜损坏，可能会影响特性。请勿强力按压或以锐利的物体弄伤特氟龙膜。
 如果校正时的指示值为零抑制值以下，则无法进行校正。
如果在校正时导入了所需的校正气体且指示值低于零抑制值，请先使用使指示值高于零抑制值的气体进行
校正，然后再使用所希望的校正气体进行校正。
 接通电源时，暖机时间（3 分钟）结束后，指示值可能因无通电时间和环境而改变。此时请等待其归零。
 SGF 传感器的保修年数与检测对象气体无关，一律 3 年。

159 / 17110. 产品规格

10-3 各种原理的检测对象气体＜半导体式传感器的暖机时间＞
无通电期间
10 分钟以内
1 小时以内24 小时以内72 小时以内
10 日以内1 个月以下1 - 3 个月以下3 个月以上暖机时间
30 分钟以上
2 小时以上4 小时以上24 小时以上
2 日以上
7 日以上
14 日以上1 个月以上160 / 17110. 产品规格

10-3 各种原理的检测对象气体

10-3-4 热线型半导体式
传感器型号

气体名称

气体
符号

F.S.

1 数位

第一

第二

校正用气

使用

使用

警报点

警报点

体

温度范围

湿度范围

SHF-8601

甲烷

CH4

5000ppm

25ppm

2000ppm

4000ppm

CH4-30 - 70℃

SHF-8601

一氧化碳

CO

1000ppm

10ppm

250ppm500ppmCO

0 - 70℃

20 - 95%RH
以下※

SHF-8602

异丁烷SHF-8603

氢

i-C4H102000ppm

10ppm

500ppm1000ppmi-C4H10-30 - 70℃

H2

2000ppm

10ppm

500ppm1000ppm

H2-30 - 70℃

保修年数

3年

（无结露）

※ 绝对湿度 46g/m3 以下（等同 37℃ 95%RH）

注记 硅气体（有机硅类气体）或各种卤素气体、硫化物气体、酸性气体的存在可能会导致传感器特性发生显著劣化。
 可选配除硅过滤器和活性碳过滤器。
 请根据使用环境的湿度制作校正用气体。
 进行校正时，请先在清洁的环境下根据无通电时间进行暖机，再进行零位校正和间距校正。暖机时间请参阅以下的＜热线型半导体式传感器的暖机时间＞。
 即使是检测对象气体以外的可燃性气体，指示值也可能会发生变动。
 指示值也可能因安装场所的环境变化（温度、湿度等）而发生变动。
 如果氧气浓度降低，指示值可能会降低。
 作为基底气体，如果同时存在高浓度的 CO2、Ar、He 等热传导率大的气体，可能会受到干扰的影响。
 如果使用抽吸法检测蒸气压低的气体，从高温侧吸入气体，在低温侧进行检测时，气体可能会雾化，导致传感器内部结露或因烧结而堵塞。使用时请注意蒸气压。
 如果在使用寿命诊断、自我诊断功能时警告显示警告，请实施维护。
 实施校正时，请使用可燃性传感器校正盖。如果不使用可燃性传感器校正盖，校正精度可能会降低。
 如果与进行零位校正或间距校正的环境温度相比，温度变化达到 ±20℃以上，则指示精度可能会降低。
此时请再次进行零位校正或间距校正。
 如果受到外部的强力冲击或振动，指示值可能会改变。此时请进行零位校正和间距校正。
 如果长时间持续接触高浓度气体，零点可能会改变。此时请进行零位校正和间距校正。
 在气体连续存在的环境下，指示值可能会降低。
 如果检测到超过 F.S 的气体，校正值可能会出现偏差。如果可能存在超过 F.S 的气体，启用传感器保护功能，即可减轻其影响。但是，传感器保护功能启用时，如果超过 F.S.，为了断开对传感器的供电，在复位
前无法进行检测。从传感器保护的状态下复位时，请在周围环境清洁的状态下实施。
如果长期（1 天以上）处于传感器保护状态，请按照保护状态的时间进行暖机。
（如果启用传感器保护功能，超标锁定功能也将启用。
）
 更换传感器时，传感器可能会处于高温状态。更换传感器时请小心避免烫伤。如果传感器很烫，请断开电
源，静置直至温度降下来。
 接通电源时，暖机时间（3 分钟）结束后，指示值可能因无通电时间和环境而改变。此时请等待其归零。
 SHF 传感器的保修年数与检测对象气体无关，一律 3 年。

161 / 17110. 产品规格

10-3 各种原理的检测对象气体＜热线型半导体式传感器的暖机时间＞
无通电期间
10 分钟以内
1 小时以内24 小时以内72 小时以内
10 日以内1 个月以下1 - 3 个月以下3 个月以上暖机时间
SHF-8603
10 分钟以上
30 分钟以上
1 小时以上4 小时以上2 日以上
7 日以上
14 日以上1 个月以上其他 SHF 传感器
1 小时以上2 小时以上4 小时以上24 小时以上
2 日以上
7 日以上
14 日以上1 个月以上※ 请注意，SHF-8603 和其他 SHF 传感器的暖机时间不同。

162 / 17110. 产品规格

10-3 各种原理的检测对象气体

10-3-5 恒电位电解式
传感器型号

气体名称

气体
符号

F.S.

1 数位

第一

第二

校正用

使用

警报点

警报点

气体

温度范围

硫化氢

H2S

100ppm

1ppm

20ppm

40ppm

H2S

-40 - 70℃※3

ESF-A24RH※1 硫化氢

H2S

30ppm

0.2ppm

10ppm

20ppm

H2S

-40 - 70℃※2

ESF-A24A

二氧化氮

NO2

15ppm

0.1ppm

5.0ppm10.0ppmNO2-40 - 70℃※3

ESF-X24P2

氧气

O2

25%

0.1%

18.0%

18.0%

N2

-40 - 70℃※3

ESF-A24R

使用湿度范围
20 - 90%RH（无结露）
40 - 95%RH（无结露）
20 - 90%RH（无结露）
20 - 90%RH
（无结露）

保修年数

3年

3年

3年

3年

※1 启用前储存在密闭容器内。本传感器支持高湿度环境。
※2 请在-25℃以下充分适应使用环境后进行调整。在 40 - 70℃的环境下累计可使用 1 星期。
※3 请在充分适应使用环境后进行调整。（但是，-25℃以下、+55℃以上要适应需耗时 1 天以上。在 55 - 70℃的环境下
累计可使用 1 星期。）

注记 如果在使用寿命诊断、自我诊断功能时显示警告，请实施维护。
 实施校正时，请使用 ESF 传感器校正盖。如果不使用 ESF 传感器校正盖，校正精度可能会降低。
 进行校正时，请在清洁的环境下进行零位校正。
 急剧的温度/湿度变化可能造成指示值变动。
 请勿对氧气计施加剧烈的压力变化。否则指示值会短时改变，无法准确测量。
 氧气计不应使用氮气以外的平衡气体。否则指示误差变大，无法准确测量。
 进行校正时，请将流量控制为 0.5 ± 0.1L/min。如果使用不同的流量进行校正，可能会降低校正精度。
 如果使用扩散式检测仪，气体灵敏度会受环境中风速的影响。
 ESF 传感器的保修年数因传感器型号而异。请参阅上表。

163 / 17111. 附录11-1 检测原理11附录
11-1 检测原理
11-1-1 非分散型红外线吸收法（红外线式）＜传感器概要＞气体检测传感器利用多种气体吸收红外线，用红外线照射测量池，检测因检测对象气体的吸收而导致的红外线变
化量。不按波长分离（分散）红外线，连接检测所有进入特定波长范围的红外线。

＜传感器的结构和原理＞
 结构
在红外线光源和红外线传感器之间配置测量池和光学过滤器。从红外线光源发出的红外线通过测量池内部和光
学过滤器，被红外线传感器检测到。光学过滤器选择性地透过检测对象气体吸收的红外线波段。
 原理
检测对象气体导入测量池内部后被红外线吸收，因此红外线传感器检测到的红外线量减少。利用这一原理，导
入几种已知浓度的检测对象气体，求红外线减少量与检测对象气体浓度的关系（标定曲线）。使用该标定曲线，导入浓度未知的检测对象气体时，根据所获得的红外线减少量求气体浓度。
＜结构图＞＜标定曲线（一例）＞

反射镜

检测对象气体检测对象气体红外线的减少量

红外线

红外线

光学过滤器
检测对象气体的浓度红外线光源红外线传感器164 / 17111. 附录11-1 检测原理

11-1-2 新型陶瓷式＜传感器概要＞使用超微粒化的氧化催化剂（新型陶瓷式），检测从低浓度(ppm)到爆炸下限(LEL)的大范围的气体浓度。
是本公司自主开发的划时代可燃性气体专用的气体检测传感器。

＜传感器的结构和原理＞
 结构新型陶瓷式传感器由检测元件和补偿元件构成（部分传感器无补偿元件）。检测元件通过在贵金属线圈上烧结对
可燃性气体呈活性的超微粒化氧化催化剂（新型陶瓷）后构成，因检测对象气体产生燃烧反应。
补偿元件通过在贵金属线圈上烧结对可燃性气体呈惰性的物质后构成，对环境的影响进行补偿。
 原理
在被贵金属线圈加热到 300 - 450℃的检测元件表面，可燃性气体燃烧后，元件的温度上升。随着这种温度变化，构成元件的贵金属线圈的电阻值也会发生变化。电阻值变化基本与气体浓度成正比。利用桥接电路提取该电阻值的变化量作为电压，求出气体浓度。
＜传感器元件示意图＞
补偿元件・・・对环境的影响进行补偿

检测元件・・・检测气体
氧化催化剂+
氧化铝担体

玻璃＋
氧化铝担体

贵金属线圈

贵金属线圈

0.5 - 1.4mm

0.5 - 1.4mm
0.8 - 1.4 mm

0.8 - 1.4mm＜桥接电路＞
元件温度降低⇒电阻减少补偿元件

元件温度上升
⇒电阻增加

＋
V
－

侧电阻 R
传感器电压 Vs

165 / 171

检测元件
传感器输出

11. 附录11-1 检测原理

11-1-3 半导体式＜传感器概要＞金属氧化物接触检测对象气体时会引发电阻值变化，本方法以电阻值变化作为气体浓度。此气体检测传感器属于
通用型，可快速应对从毒性气体到可燃性气体的所有气体检测。

＜传感器的结构和原理＞
 结构
由加热线圈和氧化铝管上形成的金属氧化物半导体(SnO2)构成，氧化铝管的两端有用于测量半导体电阻的 2 个
金电极。
 原理
在被加热线圈加热到 350 - 400℃的金属氧化物半导体表面上，大气中的氧气以 O-或 O2-的形态吸附在表面上，半导体保持恒定的电阻值。如果甲烷气体接触或吸附在表面上，氧气就会被吸附的 O2-离子氧化分离。
此时，传感器表面产生如下反应。
CH4 ＋ 4O2- → CO2 ＋ 2H2O ＋ 8e换言之，甲烷气体吸附在传感器表面争夺吸附的氧气，会使传感器内部的自由电子增加，因此导致电阻值降低。
通过测量这种电阻值的变化，求出气体浓度。
＜传感器元件示意图＞
金属氧化物半导体金电极＜驱动电路＞
零位校正器

直流电源

检测元件

信号放大器

加热器线圈
氧化铝管

引线

加热器
电压
调整器

H2O

H2

负载电阻

H2 ＋ O2- → H2O ＋ 2e-

166 / 171

指示计

间距校正器

11. 附录11-1 检测原理

11-1-4 热线型半导体式＜传感器概要＞白金线圈与接触气体时电阻值会发生变化的金属氧化物半导体结合，本方法检测白金线圈的电阻变化作为气体浓度。该传感器用于检测高灵敏度的低浓度检测气体，仅由在白金线圈周围烧结金属氧化物半导体的检测元件构成。

＜传感器的结构和原理＞
 结构仅由在白金线圈周围烧结金属氧化物半导体的检测元件构成。
 原理
检测元件的电阻值(R)是半导体电阻值(RS)和白金线圈电阻值(RH)的合成电阻。
被白金线圈的加热器加热到 300 - 400℃的检测元件保持恒定的电阻值。甲烷气体等一旦接触到检测元件，由于
吸附于金属氧化物半导体表面的氧气分离，使得可以在半导体内部自由移动的电子数量增加，半导体电阻值降低。其结果导致检测元件整体的电阻值也会降低。通过利用电压等提取该电阻值的变化量，求出气体浓度。

＜检测元件示意图＞

金属氧化物半导体＜驱动电路＞白金线圈
0.5 - 1.4mm

恒定电流源

0.8 - 1.4mm

167 / 171

检测元件
半导体材料
⇒电阻变化
⇒传感器输出指示计11. 附录11-1 检测原理

11-1-5 恒电位电解式＜传感器概要＞检测对象气体在保持恒定电位的电极上被电解，本方法检测此时产生的电流作为气体浓度。这种气体检测传感器
检测毒性气体最有效，可以通过选择设定电位选择性地检测气体。

＜传感器的结构和原理＞
 结构在透气膜上配置放有贵金属催化剂的电极(工作电极)、参比电极和对电极，将这些电极装入充满电解液的塑料容器中。
 原理
使用恒电位仪电路，使得工作电极和参比电极之间保持恒定的电位。检测对象气体穿过透气膜到达工作电极，
在那里发生电解反应。此外，会发生副作用，使得在对电极上流过与工作电极同量的电流。
例） 使用硫化氢(H2S)时，发生以下反应。
工作电极 ：H2S+4H2O → H2SO4+8H++8e（e-为电子）对电极
：2O2+8H++8e- → 4H2O
此时发生的电流与气体浓度成正比，因此通过测量流向工作电极的电流可以求出气体浓度。
＜结构图＞
恒电位仪电路参比电极工作电极

检测对象气体

电解液

气体渗透膜

催化剂

168 / 171对电极容器

11. 附录11-2 术语的定义

11-2 术语的定义
术语

定义

初始清空

此功能用于在通电后在短暂的时间内不发出警报。
刚通电后，检测仪的输出短时会发生变动，因此避免发出警报。

全量程检测范围的最大值。

校正

使用校正气体确认仪器指示值的行为。
也包括调整仪器指示值以匹配校正气体浓度值的行为。

零抑制

本功能隐藏（抑制）低于测量下限值的指示值变动，意图忽略因温湿度变化、噪声影响等非气体引发的传感器输出变化。

警报延迟时间

为防止外部侵入的噪声导致误警报，本功能在指示值达到警报点时不立即发出警报，而是暂时让警报运行引而不发。

INHIBIT

本功能在维护仪器等情况下暂时停止气体检测。也被称为“点跳过”。

vol%一种单位，以百分率表示在体积中含有多少对象气体。

ppm

一种单位，以 100 万分之 1 (part per million)表示在体积中含有多少对象气体。

%LEL

在可燃性气体中，将对象气体的爆炸下限浓度(Lower Explosive Limit)设为 100%
时的单位。（例：如果在体积中含有 5%甲烷，则其爆炸下限浓度为
100%LEL。）

169 / 17111. 附录产品保修产品保修
1. 自购买之日起 3 年内，如果按照使用说明书/主机上粘贴标签中的注意事项所述的正常使用条件下发生了故障，本公司将免费维修。
2. 维修、维护等售后服务，请咨询总公司营业部或就近的营业网点。
3. 如需异地出差维修，本公司将收取实际的差旅费。
4. 即使在保修期内，以下情况也需有偿维修。
(a) 由于错误使用和不当修理或改装而导致的故障和损坏。
(b) 由本公司以外或本公司指定服务机构以外的任何人维修或改装而导致的故障和损坏。
(c) 购买后因安装场所搬迁、运输、倾倒、坠落、储存方面的不当而导致的故障和损坏。
(d) 因火灾、地震、洪水、雷击、其他自然灾害、污染、电压异常、电源（电压、频率）超出额定值等外部因素导致的故障和损坏。
(e) 故障由本产品以外的原因引起时。
(f) 更换易耗品（电池、保险丝等）。

传感器保修
1. 自购买之日起或传感器有偿更换之日起 3 年内（部分传感器除外），如果按照使用说明书/主机上粘贴标签中的注
意事项所述的正常使用条件下发生了故障，本公司将免费更换。各型号传感器的保修年数，
请参阅“10-3 各种原理的检测对象气体”。
但是，自购买之日或传感器有偿更换之日起，每年接受至少 1 次检查是传感器保修的前提条件。
2. 维修、维护等售后服务，请咨询总公司营业部或就近的营业网点。
3. 如需异地出差更换，本公司将收取实际的差旅费。
4. 即使在保修期内，以下情况也需有偿更换。
(A) 使用时没有遵守使用说明书/主机粘贴标签中的注意事项。
(b) 由本公司以外或本公司指定服务机构以外的任何人维修或改装而导致的故障和损坏。
(c) 购买后因安装场所搬迁、运输、倾倒、坠落、储存方面的不当而导致的故障和损坏。
(d) 因火灾、地震、洪水、雷击、其他自然灾害、污染、电压异常、电源（电压、频率）超出额定值等外部因素导致的故障和损坏。
(e) 故障由本产品以外的原因引起时。
(f) 在安装环境中因共存气体（酸性气体、碱性气体、醇类、硅气等）的影响而导致传感器气体灵敏度的变化、
故障和损坏。

170 / 17111. 附录

修订记录

修订记录
版 次
0

修订内容

发行日期
2023.7.14

初版(PT2-2872/PT2cn-2870)

171 / 17111. 附录修订记录

172 / 17111. 附录修订记录

173 / 17111. 附录修订记录

174 / 17111. 附录修订记录

175 / 17111. 附录修订记录

176 / 17111. 附录修订记录

177 / 17111. 附录修订记录

178 / 17111. 附录修订记录

179 / 17111. 附录修订记录

180 / 17111. 附录修订记录

181 / 17111. 附录修订记录

182 / 17111. 附录修订记录

183 / 17111. 附录修订记录

184 / 17111. 附录修订记录

185 / 17111. 附录修订记录

186 / 17111. 附录修订记录

187 / 17111. 附录修订记录

188 / 17111. 附录修订记录

189 / 171