INSTRUCTION MANUAL

ModelNo.

SCG-***HOLE CHAMFER MEASUREMENT

S-LINE INTERNAL CHAMFER GAUGE
Thank you for purchasing the S-LINE CHAMFER GAUGE.
This precision instrument performs a simple yet highly accurate measurement of chamfer maximum diameter.
● For safe and proper use of this product, please read this instruction manual before use and follow the procedures
described. Please keep manual where it is accessible to user for future reference.
● Keep this manual with the instrument if transferred or leased to a third party.
● For inquiries about this product, please contact dealer or Niigata Seiki at the address listed on the following page.

■PART IDENTIFICATIONSAFETY※ Model SCG-4090 shown

20100 190

304047080

38 20 223624342632 30 2890 100 1101150

0.01 mm605160

2:150

Indicates risk of personal injury orCAUTION: property
damage if not followed.

1801701401301206※Can not be removed

7or replaced.

2. Limit Markers
3. Pointer4. Short Hand
5. Bezel
6. Gauge Mount
7. Shaft
8. Meas. Head
9. Ref. Surface
10. Meas. Taper8910

hardened steel and sensitive to corrosion. Pleasekeep well protected with rust inhibitor when not inuse.
● Only use with the dedicated Dial Gauge. Use of
any other gauge will cause reading error.
● Do not remove Dial Gauge, or loosen the holding
screw. Loosening screw will cause reading
inaccuracy.
● Use care when contacting Measuring Head toworkpiece and avoid impact; there is risk ofdamage to Measurement Taper.
● Instrument is designed for measuring the ID
chamfer max. diameter.
Use for any other purpose may cause accident orinjury.

APPLICATIONS & FEATURES● The chamfer diameter (max diameter) for arange of chamfer angles can be measured.
● Dedicated Dial Gauge for direct reading ofmeasurement.

SPECIFICATIONModel
SCG−2090
SCG−4090
SCG−20127
SCG−40127

127°

180
170

2:1
0.01 mm60
7080

38 20 223624342632 30 2890 100 110

160150140130120

Push in until there
is no gap

Diam.

（Fig.２）

ZERO ADJUSTMENT※ Zero point is set at factory, but we recommend
periodic adjustment as required.

Setting Zero Point on Flat Reference Surface① A smooth, hard surface is required as Reference
Surface.
② Place the Measurement Head perpendicular tothe Flat Reference Surface and slowly push in
until there is no gap. (Fig 3)③ The number printed on the side of the DedicatedGauge is the Reference Value used for zeroadjustment. (Fig. 4)
Adjust the reading to match the Reference Valueby rotating the Bezel of the Dedicated Gauge.
(Fig. 5)
When the Gauge Reading matches the Reference
Value, the zero adjustment is complete.
※ Repeat the measurement several times toconfirm that the value is stable.

Meas. Head90°Reference
Surface（Fig.３）
Value−

Value＋Ref.
Value

20100 190

3040

2:1

500.01 mm

607080（Fig.５）（Fig.４）180
170

38 20 223624342632 30 28

90 100 110

160150140130120Bezel

Setting Zero Point Using a Reference Standard① Prepare an ID Chamfer Reference Standard for use.
② Hold the Measurement Head perpendicular to the Reference Standard and slowly push in until
there is no gap between the Reference Standard and the Measurement Head Reference Surface.
③ Adjust the reading to match the Reference Standard by rotating the Bezel of the Dedicated Gauge.
(Fig. 5) When the Gauge Reading matches the Reference Standard, the Zero Adjustment is complete.
※ Repeat the measurement several times to confirm that the Value is stable.

AFTER USE CARE, STORAGE● Clean outside surfaces regularly using a cloth to remove any dirt or contamination.
● After use, place in a rust preventive bag or use a corrosion inhibiting oil for protection.

● If surfaces are dirty, wipe with a clean cloth moistened with a cleaner such as Niigata Seiki STC-500.

Measurement Usable Chamfer
Taper Angle Angle Range
90°

0 190

50

① Hold the Measurement Head perpendicular
to the workpiece and slowly push in until
there is no gap between the workpiece （Fig.１）
and the Measurement Head Reference
Meas. Taper
Surface. (Fig. 1)② The Dial Gauge will indicate the maximum
diameter of the chamfer. (Fig. 2)

1040

19,58

1. Dedicated Dial Gauge● Measurement Head and Measurement Taper are90°

2030

Press

Always follow the procedures specified below in order toprevent harm to yourself or others, and to prevent damage toproperty.

23

PLEASE OBSERVEMEASUREMENT

Meas. Head90°or Less
90°〜120°

Range(mm)0.5〜2020〜400.5〜20
20〜40

Graduations(mm)

Accuracy
(mm)

340

0.01
±0.0150.02

Weight(mm)640340640● Store in location protected from direct sunlight and humidity, and accessible only to authorized personnel.

REPAIRS AND SERVICE● If Gauge does not operate properly, or if you have anyquestions, please contact distributor or place of purchase.

● Please note, manufacturer is unable to respond to inquires orprovide service directly. Please contact distributor or place ofpurchase.

Niigata seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

取扱説明書品番SCG-***

Sライン内径面取りゲージ

面取り加工穴測定器

この度は「Sライン内径面取りゲージ」をお買上げいただきありがとうございます。
この商品は内径面取り及びテーパ部の最大径を、簡単・高精度に測定することが可能な精密測定器です。
●正しく安全にお使いいただくため、ご使用の前に本取扱説明書を必ず読み、記載の手順に従ってご使用ください。
お読みになった後は、お使いになる方がいつでも見ることができる場所に大切に保管してください。
●第三者に譲渡・貸与される場合も、この説明書を必ず添付してください。
●当商品に関するお問い合わせは、お買い求めの販売店もしくは当社にご連絡ください。

安全上のご注意■各部の名称 ※イラストはSCG-4090

必ずお守りください2203100 19030404708038 20 223624342632 30 28

90 100 110

11500.01 mm

6051602:150『傷害を負う、または物注意 お守りいただかないと的損害が発生するおそれがある内容』です。

180
170

1401301206※交換・取り外し不可

2. リミット針
3. 長針
4. 短針
5. 外枠
6. ダイヤルゲージ
ホルダ
7. 外筒
8. 測定ヘッド
9. 測定基準面
10. 測定テーパ部

78910

用途・特長

●内径C面取部及び任意角度の面取部の最大径を測定できます。
●測定値は、専用ダイヤルゲージで直接読み取ることができます。

製品仕様
品

番

SCG−2090
SCG−4090
SCG−20127
SCG−40127

測定テーパ部
角度

使用可能
面取角度範囲

測定範囲

90°

90°
以下

0.5〜2090°〜120°

0.5〜20127°

（mm）

20〜40
20〜40

目量（mm）

精度（mm）

340

0.01±0.0150.02質量（g）

6403406400 190

40180170
1602:150①測定テーパ部を測定物に直角に当て、測定物と基準平面の間に隙間が無くなるまで、測定ヘッドを押し込んでください。 （図１）（図１）
測定テーパ部②専用ダイヤルゲージの示す測定値が、内径面取部の最大径となります。
（図２）

10150

0.01 mm

60708038 20 223624342632 30 28

90 100 110

140130120

隙間が無くなるまで押し込む

測定値

（図２）

ゼロ点の調整※出荷時にゼロ点を調整済ですが、使用頻度に合わせて調整をお勧め致します。

基準平面でゼロ点を合わせる場合

直角①平滑で硬質な基準平面をご用意ください。

②測定テーパ部先端を基準平面に直角に当て、測定基準面と基準平面の間に隙間が無くなるまで、測定ヘッドを押し込んでください。
（図３）

基準平面

（図３）

③専用ダイヤルゲージ側面に記載されている数値が、基準平面で合わせる場合の基準値です。
（図４）
この基準値に合うよう、専用ダイヤルゲージの外基準値枠を回転させて、測定値を調整してください。
（図５）
測定値と基準値のサイズが一致したら、ゼロ点が合っています。
※ゼロ点調整は数回繰返し、数値が安定していることを確認してください。

測定ヘッド

数値を＋20

数値を−100 190

304019,581. 専用ダイヤルゲージ●測定ヘッド・測定テーパ部は焼入れ鋼であるため、使用後は必ず防錆処理を行ってください。
●専用ダイヤルゲージ以外のダイヤルゲージをご使用になると、正確な数値が表示されませんので、使用しないでください。
●ダイヤルゲージホルダのねじをゆるめたり、専用ダイヤルゲージを取り外したりしないでください。測定値・精度に影響が出ます。
●測定器をワークに当てる際は、強い衝撃を与えないようワークに対してゆっくりと測定器を当ててください。測定テーパ部を破損させる危険があります。
●本製品は内径面取り及びテーパ部の最大径測定器です。用途以外のご使用は、事故やけがの原因となりますので絶対におやめください。

直角2030

押し込む

お使いになる人や他の人への危害、財産への損害を未然に防止するため、必ずお守りいただくことを下記のように説明しています。

測定方法

測定ヘッド

2:1

500.01 mm

607080（図４）18017038 20 2236243426
32 30 28

90 100 110

160150140130120（図５）

外枠

お手持ちの基準マスタでゼロ点を合わせる場合①内径面取用の基準マスタをご用意ください。

②測定テーパ部を基準マスタに直角に当て、測定基準面と基準マスタの間に隙間が無くなるまで、測定ヘッドを押し込んでください。
③基準マスタのサイズに合うよう、専用ダイヤルゲージの外枠を回転させて、測定値を調整してください。
（図５） 測定値と基準マスタのサイズが一致したら、ゼロ点が合っています。
※ゼロ点調整は数回繰返し、数値が安定していることを確認してください。

使用後のお手入れ・保管方法●外部から触れることのできる研磨部品は、定期的に清潔な布で拭いて汚れを除去してください。
●汚れがひどい場合は別売の工具クリーナ STC-500 等を使用して清潔な布でふき取ってください。
●防錆油を薄く塗るか、防錆袋に入れるなどの防錆処理を行ってください。
●直射日光や湿気を避け、管理者以外が触れない状態で保管してください。

修理の際は●正常に作動しない場合や、不明な点がございましたらお買い上げの販売店または、発売元までご連絡ください。
●お問い合わせや、ご連絡が無いまま直接発売元に修理品などを送付されても処理、対応ができない場合がありますのでご了承ください。

発売元〒955-0055 新潟県三条市塚野目5丁目3番14号(0256)33-5502(代) FAX(0256)33-5528

URL http://www.niigataseiki.co.jp

I411-K2003