§ PLUG-IN TRANSDUCER §■ PLUG-IN 1 OUTPUT TYPE
ProductCOMPENDIUM
Signal/Sensor/AC transducer

Type codeOutlinesConverts a DC input signal into a unified signal which wasisolated. Response time 0.5s/99%
Converts a DC input signal into a unified signal which wasisolated. Response time≦500μs/90%
Outputs a pulse input signal through an isolated 2-output
relay contact or an isolated open collector.
DC signal V-Ⅰconversion. Input and output not insulated.
Supplies electric power to a 2-wire transmitter insulates andconverts signal from the transmitter into a proportional DC
signal.
Supplies electric power to a two-wire transmitter, converts
signal (4-20mA) from the transmitter into 1-5V by a precise
resistance.
Converts a nonlinear DC signal into a linearized output
signal.
Converts a DC signal into another one which was proportional
to the square of the former one.
Converts a DC signal into another one which was proportional
to the square root of the former one.
Outputs a DC signal which is equivalent to multiplication of
two DC signals.
Outputs a DC signal which is equivalent to quotient of two DC
signals.
Sets upper/lower limit for the proportional outputs.
Outputs a DC signal which is equivalent to sum of two DCsignals.
Output converts at a constant speed being set.
Holds output when "HOLD" terminal is turned OFF.
A transducer that has a limiter (fixed) function only for thelower limit.
Reverses gradients of input/output signal, then outputs them.

Isolator

TP2-□□

Ultrahigh speed isolator

HSTP1 -□□

Pulse isolator
DC transducer
Insulation type
distributor

PPTP2-□□TP-

DTP2-C1□□DTP-C10□

Linearizer

LTP1-

Square transducer

SQTP1 -□□

Square root extraction
transducer
Multiplying transducer

SRTP1 -□□

MTP1-

□□

Dividing transducer

DITP1-□□

Analog limiter
Adding transducer

ALTP ADTP1 -□□□□Constant response
Analog memory
Isolator with lowerlimiter
Reverse isolator

CRTP AMTP TP2-

□□
□□
□□L

RVTP2-□□

VFTP2-□□□UGTP2-□□□

PRTP2-□□□

HTP1-□□□□□□

Distributor

Analog pulse transducer
Ultraslow pulse
transducer
Pulse rate transducer
Thermoelectric
temperature transducer
Resistance temperaturetransducer
Potentiometertransducer
Revolution-speedtransducer (Frequency
proportion type)
Revolution-speedtransducer (AC voltage
proportion type)Selsyn transducer
AC current transducer
AC voltage transducer
Frequency transducer
AC current transducer
AC voltage transducer
AC current transducer
AC voltage transducer
DC power transducer

Products List□□

RTP2-□□□

GTP2-□□□□

Outputs a pulse which frequency was proportional to a DC
input signal.
Outputs a DC signal which was proportional to frequency ofan ultraslow pulse.
It converts a pulse signal into another pulse signal which
frequency was divided by n.
Converts thermal electromotive force of a thermocouple into aDC signal which was proportional to temperature.
Converts resistance of a three-wire thermal resistance into aDC signal which was proportional to temperature.
Outputs a DC signal which was proportional to resistance of apotentiometer.
Outputs a DC signal which was proportional toRevolution-speed (frequency) of a tacho generator.

GVTP2-□□□□

Outputs a DC signal which was proportional toRevolution-speed (voltage) of a tacho generator.

STP1-□□□

AETP2-□□□

VETP2-□□□

FTP2-□□□

RHTP2-□ATP2-□□

VTP2-□□

AP2-□□

VP2-□□□□

DWP1 -□□□□□

Converts displacement of a Revolution angle of a selsyn
transmitter into a DC signal.
Outputs a DC signal which was proportional to RMS value ofan AC current input.
Outputs a DC signal which was proportional to RMS value ofan AC voltage input.
Outputs a DC signal which was proportional to frequency.
Outputs a DC signal which was proportional to AC current.
Power-free Constant current output
Outputs a DC signal which was proportional to AC voltage.
Power-free Constant current outputOutputs a DC signal which was proportional to AC current.
Power-free Load fixed type.
Outputs a DC signal which was proportional to AC voltage.
Power-free Load fixed type.
Outputs a DC signal which was proportional to DC power.

Withstandvoltage
AC2000V
AC1500V
AC2000V
AC2000V

AC1500V
AC1500V
AC1500V
AC1500V
AC1500V
AC1500V
AC2000V
AC2000V
AC2000V
AC2000V
AC2000V
AC1500V
AC2000V
AC2000V
AC2000V

AC2000V
AC1500V
AC2000V
AC2000V
AC2000V
AC2000V
AC2000V
AC2000V
AC2000V
AC1500V

1§ PLUG-IN TRANSDUCER §COMPENDIUM■ PLUG-IN 2-OUTPUT TYPE Signal/Sensor/AC transducer
Product

Type codeIsolator
Distributor
Distributor withsquare root
extraction
Thermoelectric
temperaturetransducer
Resistancetemperaturetransducer
Potentiometertransducer
AC current
transducer
AC voltage
transducer
Frequencytransducer

Outlines

WHTP2 -□□□□□

Converts a DC input signal into a unified signal which wasisolated.
Supplies electric power to a 2-wire transmitter and converts
signal from the transmitter into a proportional DC signal.
Supplies electric power to a 2-wire transmitter and converts
signal from the transmitter into a DC signal which was
proportional to the square root of the signal.
Converts thermal electromotive force of a thermocouple into aDC signal which was proportional to temperature.

WRHTP2-□□□□□

Converts resistance of a 3-wire thermal resistance into a DCsignal which was proportional to temperature.

WTP2-WDTP2-C7□□□□□□WSRDTP2-C7□□□

WRTP2-Z□□□

WAETP2 -□□□□
WVETP2 -□□□□
WFTP2 -□□□□□

Outputs a DC signal which was proportional to resistance of apotentiometer.
Outputs a DC signal which was proportional to RMS value ofan AC current input.
Outputs a DC signal which was proportional to RMS value ofan AC voltage input.
Outputs a DC signal which was proportional to frequency.

Withstandvoltage
AC2000V
AC2000V
AC2000V

AC2000V

AC2000V
AC2000V
AC2000V
AC2000V
AC2000V■ Soft spec type
ProductType code

Adding/subtracting
transducer

CADTP1 -□□Multiplying/dividing
transducer

CMLTP1 -□□Temperature/pressure
correcting transducer

CLTP1-□□

Function generatingtransducer
Analog backup
transducer

CFGTP1 -□□CAMTP1 -□□CVFTP1 -□□Voltage pulse transducer

Outlines
Does adding and subtracting of three inputs, and then outputs
a DC signal equivalent to the value.
Parameters can be changed by CCM-1.
Does multiplication and division of three inputs, and thenoutputs a DC signal equivalent to the value.
Parameters can be changed by CCM-1.
Processes temperature/pressure condition and converts it intoa DC signal which was proportional to flow rate.
Parameters can be changed by CCM-1.
Does broken line operation of a DC input 15 polygonal lines
maximum. Parameters can be changed by CCM-1
Provides output with a backup when a computer or a PID
controller was down. Follow-up movement and output backup
are settable by CCM-1.
Outputs a pulse of frequency which was proportional to a DCinput. Pulse frequency, pulse width and output cut against alow input are settable by CCM-1.

Withstandvoltage
AC1500V

AC1500V

AC1500V
AC1500V
AC1500V

AC1500V■ Alarm setter
Product

Type code

Outlines

Alarm setter(digital % scale)
Alarm setter(actual scale)SDD-□-105□-□SD-□-105□-□

Alarm setter(LCD)

SDLC-105-□

Deviation alarm
setter (LCD)

SDDV-105-□

AC voltage alarmsetter

Products List

SVD-□-105□-□□Compares a preset value of digital % scale with a direct input
signal, and outputs a contact signal.
Compares value of an actual scale setter with a direct input
signal, and outputs a contact signal.
Compares a preset value with a direct input signal, and thenoutputs a contact signal. 4 digit LCD indication. Actual scaleindication is settable.
Compares deviation between two DC signal inputs anddeviation of each input with a preset value, and then outputs acontact signal.
Inputs AC voltage and outputs a contact signal

Withstandvoltage
AC1500V
AC1500V
AC2000V

AC2000V
AC2000V

2§ PLUG-IN TRANSDUCER §COMMON STANDARD SPECIFICATION/TYPE CODE DESIGNATION■Common standard specifications●High quality/high reliability
Highly reliable electronic parts are adopted.
Aging tests of each part as well as burn-in aging test of the product under a high temperature areimplemented.
●PCB treatment
In order to reinforce insulation resistance stability of PCB surfaces and prevent the surfaces frominsulation deterioration, B side of the PCB was cleaned and coated with high humidity resistant varnish
after parts installation.
●Output limiter circuit
Even if an excessive input is applied, the product confines the output to about 1.5 times of rating andprotects the output side equipments.
ItemSpecification

Tolerance% against output span

Influence of temperature

23±10℃ tolerance %Influence of frequency

45-65Hz tolerance %Characteristics

In conformity with JIS C 1111-1989 in tolerance(Reference) IEC, rated Hz±10% tolerance %Time it takes to fall within ±1% of the final steady-state when applied a stepped input.

Response timeStandard : ≦1.0. sec. (Insulation transducer only: ≦0.5 sec.)

Output ripple≦1%p-p against output span

External adjustment of±5% adjustable

outputAuxiliary supply

AC100V or AC200V ±10% (50, 60Hz) (DC100/110V is manufacturable only for TP2)

Overvoltage

Input

2 times (10 sec.), 1.2 times (continuity) of rated voltage

Aux.supply

1.5 times (10 sec.), 1.1 times (continuity) of rated voltage
40 times (1 sec.), 20 times (4 sec.) 10 times (16 sec.),

AC transducer

Over current

1, 2 times (continuity) of rated current

Signal transducer

10 times (5 sec.)

Between input terminal, output terminal, (auxiliary supply terminal) and outer case (earth)Insulation resistance≧50MΩ at DC500V. Non-insulation type: input terminal and output terminal conducted.

Material of outer caseFire-retardant ABS resin

Appearance color
Operating temperature/Outer caseBlack (N 1.5)

Rating plate

Dark blue (5PB 2/6)

-10-＋55℃, 30-85%RH

humidity range
Storage temperature-40-＋70℃range■Type code designation

1 output type
●Signal transducer(1)

Products List

P

(2)

–(3)

(4)

(5)

3

§ PLUG-IN TRANSDUCER §COMMON STANDARD SPECIFICATION/TYPE CODE DESIGNATION(1) Product (kind of conversion)
Mark

Product (kind of conversion)

TMark

Product (kind of conversion)

Isolator

CRT

Constant response (constant speed response)

Ultrahigh speed isolator

AMTAnalog memory

DTDistributor

T***L

Isolator with lower limiter

LTLinearizer

RVT

Reverse isolator

SQT

SquareVFT

Analog pulse

RSignal switch

HST

SRT

Square root extraction

MTMultiplying

UGT

Ultraslow pulse

DIT

Dividing

PRT

Pulse rate

ALT

Analog limiter

PPT

Pulse isolator (2-output)

ADTAdding(2) Dielectric strength voltage(3) / (4) / (5) Specification code

MarkDielectric strength voltage (50/60Hz)

None

Non-insulation

Input/output/auxiliary supply

1

AC1,500V between input/output, for 1 min.

2

AC2,000V between input/output, for 1 min.

●Sensor transducer

(1)

TP

(2)

–(3)

(4)

(5)

(1) Product (kind of input)
Mark(2) Dielectric strength voltage

Product (kind of input)

H

Thermoelectric temperature

RH

Resistance temperature

R

Potentiometer

Mark

G

Revolution-speed (Frequency proportion)

GV

Revolution-speed (Voltage proportion)

SSelsynSH

Thermoelectric alarm

SRH

Platinum alarm

Dielectric strength voltage (50/60Hz)

1

AC1,500V between input/output, for 1 min.

2

AC2,000V between input/output, for 1 min.

(3) / (4) / (5) Specification code
Kind of thermocouple, input, output, power source
Kind of thermal resistance, input, output, power source
Input, output, powerInput, normal operating voltage, output, power source●AC transducer

(1)

P2–

(2)

(3)

(4)

(1) Product (kind of input)

Dielectric strength voltage

MarkProduct (kind of input)

VAC voltage (with waveform compensation, load fixation, need no power source)

VTAC voltage (with waveform compensation, need no power source)

VET

AC voltage (RMS value)A

AC current (with waveform compensation, load fixation, need no power source)AT

AC current (with waveform compensation, need no power source)

AET

AC current (RMS value)

FT

Frequency

AC2,000V(50/60Hz) between
input/output, for 1 min.

(2) / (3) / (4) Specification code
Input, output, power●DC power transducer

DWP1 – (1)

(2)

(3)

(4)

(5)

Dielectric strength voltage
AC1,500V(50/60Hz) between input/output, for 1 min.

(1) / (2) / (3) / (4) / (5) Specification code
Electric power, input 1, input 2, output, power source

Products List4§ PLUG-IN TRANSDUCER §COMMON STANDARD SPECIFICATION/TYPE CODE DESIGNATION

2-output type
●Signal transducer

W

(1)

P

(2)

–(3)

(4)

Product (kind of conversion)T(6)(2) Dielectric strength voltage(1)Product (kind of conversion)
Mark

(5)

Isolator

DT

Distributor

SRDT

Square root extraction distributor

Mark

Dielectric strength voltage (50/60Hz)

2

AC2,000V between input/output, for 1 min.

(3) / (4) / (5) / (6) Specification code
Input, output, power source●Sensor transducer

W

(1)

P

(2)

-(3)

(4)

(5)

(6)

(7)

(1)Product (kind of input)
Mark(3) / (4) / (5) / (6) / (7) Specification code

Product (kind of input)

HT

Thermoelectric temperature

RHT

Resistance temperature

RT

Potentiometer

Kind of thermocouple, input, output, power source
Kind of thermal resistance, input, output, power source
Input, output, power
Input, normal operating voltage, output, power source(2)Dielectric strength voltage
Mark2

Dielectric strength voltage (50/60Hz)
AC2,000V between input/output, for 1 min.

●AC transducer

W

(1)

P

(2)

–(3)

(4)

(5)

Product (kind of conversion)

VET

(7)

(2) Dielectric strength voltage(1)Product (kind of input)
Mark

(6)

AC voltage (RMS value)

AET

AC current (RMS value)

FTFrequency

Mark

Dielectric strength voltage (50/60Hz)

2

AC2,000V between input/output, for 1 min.

(3) / (4) / (5) / (6) / (7) Specification code
Input, rated voltage (current), output, power source

Soft spec. type
●Signal transducerC(1)

P(2)

–(3)

(4)

(1)Product (kind of conversion)
Mark

Product (kind of conversion)

ADT

Adding/subtracting

MLT

Multiplying/dividingLT

Temperature/pressure correction

FGTFunction generating

AMTAnalog backup

VFTVoltage pulse(5)(2) Dielectric strength voltage
Mark

Dielectric strength voltage (50/60Hz)

1

AC1,500V between input/output, for 1 min.

(3) / (4) / (5) Specification code
Input, output, power source●Programming unit
CCM-1
Products List5§ PLUG-IN TRANSDUCER §COMMON STANDARD SPECIFICATION/TYPE CODE DESIGNATION

Alarm setter(1)−(2)− 105(3)(1) Scale
MarkScaleSD

Actual scale

SDD

Digital % scale−(4)(5)(2) Setting
MarkSetting

HL

Upper/lower limit setting

HH

Upper/upper limit setting

LLLower/lower limit setting

H

Upper limit setting

LLower limit setting(3) Option
MarkOption

No markStandard

DWith contact delay circuit(4) Input(5) Control power source

Digital LCD type
SDLC – 105 − (1)(2)(1) Input(2) Control power source

Deviation alarm setter

Pulse isolator

SDDV 105 − (1)PPTP2− (1)

(2)

(2)

(1) Input(1) Input(2) Control power source(2) Control power source

Power source arrester

Signal arrester

AR − (1)DA − (1)(1) Rated line voltage(1) Product (kind of conversion)

Mark
100200

Rated line voltage
AC100/110V
AC200/220V

DA − 1 (1)Kind of power source rating
Mark
1
2
3

Mark

Product (kind of conversion)TP

DC4-20mA

HT

Thermocouple

RH

Thermal resistanceRT

Potentiometer

GTPulsePower source rating≦AC125V/DC180V≦AC250V≦DC30VDA − 2 (1)Kind of power source ratingMark
1
2Power source rating≦AC125V/DC180V≦AC250V

Products List6§ PLUG-IN TRANSDUSER §Dimension■ Dimensions (mm)Fig.1
●Main body

8-pin type

●Socket (Type code: 8PFA by OMRON)

Also compatible with the following socket:
Type code: 8PFA-W by Omron*1: accessory for thermoelectric
temperature transducer*2: diode unit for current transducer

Fig.2●Main body

8-pin type

●Socket (Type code: 8PFA by OMRON)

Also compatible with the following socket:
Type code: 8PFA-W by Omron

Fig.3●Main body

11-pin type
●Socket (Type code: 11PFA by OMRON)

Also compatible with the following socket:Type code: 11PFA-W by Omron*1: accessory for thermoelectric
temperature transducer*2: diode unit for current transducer■Multiple unit installation (mm)●Panel mounting●Rail mounting●Multiplier (DM-1)

In consideration ofradiation of heat by anatural convection of air,please establish metricspaces more than thosementioned above.

Use a DIN standard 35mm rail.

Products List1