User Manual

PROFESSIONAL

WDI-10KD WDI-0342 WDI-0558SP
WDI-0542 WDI-1058SP
DI-1058
WDI-0158SP

Dial Indicator / Comparator

Model No. WDI-10

DIAL GAUGE

Thank you for purchasing the Niigata Seiki Dial Gauge.
Used with a Magnetic Base or Indicator Stand, this gauge will show the difference in height or
position relative to a zero point set at a reference position.

SAFETY PRECAUTIONS

Always follow the procedures specified below in order to prevent harm to yourself or others, and toprevent damage to property.
■Content marked as follows indicates risk of injury or damage if not followed．

WARNING Indicates risk of personal injury or property damage if not followed.
■These symbols mark content that must be observed.

・Measuring runout for rotary shafts
・Checking vise parallelism on milling machines
・Comparing parts to a master part during inspection ・Measuring flatness of surfaces and assemblies
・Confirming machine tool feed distance・Measuring machine tools positioning accuracy

Denotes a prohibition – You MUST NOT do．

■APPLICATIONS●For safe and proper use of this product, please
read this instruction manual before use and follow
the procedures described. Please keep manual
where it is accessible to user for future reference.
●Keep this manual with the instrument if transferredor leased to a third party.
●For inquiries about this product, please contact
dealer or Niigata Seiki at the address listed on thefollowing page.

Read the manual and follow all instructions.

SAFETY NOTIFICATIONS・Use for any purpose other than measuring
may damage or wear the instrument.
Improper use may also cause accident.

③ Main Scale(Indicator Face)⑦ Stem(φ8ｍｍ)※To Use the Rear Plate with Lug,
unscrew the Standard Flat
Rear Plate screws, replace,and use screws to secure.

⑧ Spindle⑨ Contact Point (Probe)①Limit Markers ………… For marking acceptance
range for Pass/Fail testing②Pointer (Needle) ……… Shows measurement on Main
Scale
③Main Scale ……………… Main Scale, rotate Bezel (⑩)(Indicator Face)to turn④Bezel Clamp Screw … Clamps Scale to prevent
rotation (WDI-0558SP/

WDI-1058SP/WDI-0158SP only)⑤Revolution Counter … Pointer indicates Main Scale
×100⑥Rev. Counter Scale … Each division is Main Scale
×100

No Rapid Motion

No Lateral Force

PREPARATION - Mounting
Dial gauge must be securely mounted such as on a comparator stand or magnetic base.
Please follow these guidelines.

Make sure Gauge holder is rigid.

Gauge Holder

・Holder must be sufficiently secure to prevent
deflection from the weight of the Gauge.
・Holder must be rigid enough to hold Gauge andnot to lift from measurement force.
・Holder support arm should be as short as possible
to prevent deflection.
※Deflection or lifting will cause measurement error
such as origin position error and inaccuracies inmeasured reading.

⑦StemSupport ArmMinimize Length

Dial Gauge must only be attached by Stem or Rear Lug.
・Mounting of gauge by other than Stem or Lug willcause inaccuracy and product damage.

During installation, do not over-tighten the Stem.

・Excessive force on the Stem may cause Spindle to bind.

※Mounting Example

TROUBLESHOOTING①Confirm that the Contact Point and Rear Plate are tightly fastened.

■Origin position shifts during measurement.

If loose, tighten Contact Point and Rear Plate Screws.

②Attach to the Gauge Holder using the Stem or Rear Lug.

Mounting Gauge by other than Stem or Lug will cause inaccuracy and product damage.

③Confirm that Pointer and Revolution Counter movement is smooth.

Using fingertip, gently press on Contact Point to move the Spindle up and down. Motion ofPointers should be smooth. If it is not smooth make sure Stem is not clamped too tight, and adjust.
Also make sure Pointer is stable at set position.

④Make sure Spindle axis is perpendicular
to measured surface.

・Temperature changes during measurement can cause repeatability error. Please try the following
solutions:●Use in location with constant temperature.
●When taking measurements, periodically adjust zero point using a Master reference to correct
for temperature induced drift.

■Measurement is not stable, or measurement accuracy is poor.

・Contact Point may be worn.
Worn Contact Point will affect accuracy. Periodically check for wear, and replace if wear isaffecting measurement accuracy.

SPECIFICATIONS
Model No.

Right Angle
Measured Object

Measured Object

HOW TO USE - Comparison Measurements①Set Up Reference Part.

① Limit Markers

Carefully lift Spindle with fingertip, and, taking
care not to hit Spindle from the side, insert theReference Part or Master under Contact Point.

Ex.) Pass Range :−0.05 to ＋0.15mm

WDI-10KD WDI-10 DI-1058 WDI-0342 WDI-0542 WDI-0558SP WDI-1058SP WDI-0158SP

Graduation

(mm)

0.01

Measurement Range(mm)

10ScaleHysteresis Error(μm)

Measuring Force

(N)

0.01

±0-50

5

50.00110±0-100

10-100-05632

≦ 1.5

≦ 1.5

≦ 1.5

≦288

98

10

14

16

16

20

Repeatability

(μm)

20
5

Weight(g)Replacement Contact Point Part No.

Adjust the gauge mount or rotate the Bezel toset the Gauge to "0".

±0-1000.01
3(mm)1/10 Rev Indication Error (μm)
Indication Error over the Entire(μm)
Measuring Range

1 Revolution

Contact Point②Set the Origin.

10.5

150.2
350.3

3180100190

Steel Ball

Steel Ball

Carbide Ball

DI-CP

DI-CP

WDI-W

AFTER USE CARE, STORAGE③Remove Reference, and begin
measurements.

Remove Reference or Master, careful not toshock Spindle. Insert part to be measured andread the measurement off the Scale.

・Do not drop or subject to shock, do notplace under heavy objects. Damage may
cause failure or poor accuracy.

⑦Stem ……………………… For holding and mountingGauge⑧Spindle …………………… Shaft moves up and downwith measurement⑨Contact Point ………… Probe which contacts(Probe)workpiece⑩Bezel ……………………… Rotate to turn Main Scale (③)⑪Flat Rear Plate………… Standard back cover⑫Lug ………………………… Alternate mounting point⑬Mounting Hole ………… For mounting Gauge⑭Rear Plate with Lug … Accessory back cover formounting Gauge using Lug

HOW TO USE - Preparation

If Spindle (Contact Point) is not at a rightangle to surface, Gauge will not operateproperly and measurement will be inaccurate.
Always keep the Spindle axis perpendicular.
※When used to check parallelism of Milling
Machine vise, use a Magnetic Base tomount the Gauge perpendicular to surface,and move it out of the way during operation
to prevent interference.

Handle With Care.

⑬ Mounting
Hole(φ6. 5mm)

⑭Rear Platewith Lug

・Rapid motion, or lateral force may damage
Gauge and cause poor accuracy.

・Use in location contrary to the above may
cause poor accuracy, damage to theproduct, or may result in accident or injury.

[Accessories] ⑫ Lug⑥ Rev. CounterScale

Do not shock Spindle.

Use in an environment which meets thefollowing conditions:●Temperature within range of 0~40°C、
humidity 30~70% (non-condensing.)●Location with minimal dust, oil, oil mist,
and protected from direct sunlight.
●Location protected from use by
children and unauthorized people.

⑩ Bezel② Pointer(Needle)

・It may damage Gauge and cause poor
accuracy.
・If Bezel Clamp Screw is removed, internal
components may come loose and become
misaligned causing product failure.

Use only as indicator Gauge.

⑪ Flat Rear Plate⑤ Revolution
Counter

Do not disassemble or modify.

・Use of product other than as described in
the manual may cause accident.

Throughout this manual, “ ” symbol indicates
RISK OF PERSONAL INJURY OR PROPERTYDAMAGE if not followed.
The “” symbol indicates something which is
PROHIBITED, and the “” symbol Indicates
REQUIRED step or necessary condition.

④ Bezel Clamp
Screw

Denotes a requirement – You MUST do．

CAUTION■PART IDENTIFICATION, FUNCTION ※Model WDI-0558SP shown① Limit Markers

Please Observe② Pointer※Setting the Limit Markers

Limit Markers can be moved to show
acceptance range for measurements.

Remove any dust or dirt after use. ※Do not lubricate.

・Wipe any contamination from Spindle sliding surface using a dry cloth, or cloth moistenedwith alcohol.
・To clean other surfaces, wipe with a soft dry cloth, or a cloth moistened with a mild cleaner.

Check for wear of Contact Point.

・Measurement accuracy will be affected by worn Contact Point. Regularly check for wear
and replace Contact Point if worn.

Store in provided case in a cool, dark, and dry location.

HOW TO USE - Parallelism, Flatness, Runout, etc.
①Position Contact Point on surface.

Carefully lifting Spindle with fingertip, and taking care not to hit Spindle from the side, position thesurface to be measured under the Contact Point.

②Set the Origin.

Adjust the gauge mount, or rotate the Bezel to set the Gauge to "0".

③Read the scale as the measured part is moved.

Slowly move the part while monitoring the Pointer and reading the measurement.

・During storage, make sure there is no force on the Spindle (such as pushed in, or lateral force.)
・Keep away from moisture and direct sunlight, and secure from unauthorized personnel.

To maintain measurement accuracy, periodic
calibration is recommended.

Calibration interval necessary to maintain accuracy will varydepending on frequency and conditions of use. Please follow
your companies guidelines and calibrate regularly.
Outside Japan,
Please contact distributor or place of purchase to inquire
about calibration services.

Niigata seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

業務用WDI-10KD WDI-0342 WDI-0558SP
WDI-0542 WDI-1058SP
DI-1058
WDI-0158SP

ダイヤル式精密比較測定器

品 番 WDI-10

ダイヤルゲージ

この度は「ダイヤルゲージ」をお買上げ頂きありがとうございます。この商品はマグネットベースやコンパレータスタンドなどの保持具に固定して、任意の基準地点を０とした時の測定値の増減を読み取る比較測定器です。

■用途の一例

安全上のご注意

取扱説明書

・マスターを基準とした加工物の検査・工作機械の精度検査
・回転軸の振れ検査●正しく安全にお使いいただくため、ご使用の前に本取扱説明書を必ず読み、記載の手順に従ってご使用ください。
お読みになった後は、お使いになる方がいつでも見ることができる場所に、大切に保管してください。

お使いになる人や他の人への危害、財産への損害を未然に防止するため、必ずお守りいただくことを下記のように説明しています。
■表示内容を無視して、誤った使い方をした時に生じる危害や損害の程度を次の表示で区分しています。

注 意 『傷害を負う、または物的損害が発生するおそれがある内容』です。

■お守りいただく内容の種類を次の図記号で区分しています。
してはいけない内容『禁止事項』です。

・フライス盤のバイス平行確認・加工、組立における面の測定
・機械加工における送り量の確認

取扱説明書をよく読み、指示に従う。

この説明書には拡大損害が予想される事項には、禁止事項には、必ずお守り頂きたい強制事項にはマークを付けて表示しています。

精密比較測定作業にのみ使用する。

④ 止めねじ

●温度０〜４０℃、湿度３０〜７０％の結露しない
場所●埃、油、オイルミストが少なく、直射日光が当たらない場所●子どもや、使用者以外が近付かない場所
・上記に反する場所での使用は、精度不良や製品の破損、事故やけがの原因となります。

⑪ 標準裏蓋

⑩ 外枠

[付属品]

⑤ 短針

本器は大切に取り扱う。

・落下などの衝撃を与えたり、重量物を乗せたりしないでください。精度不具合や故障、破損の原因となります。

⑫ 耳金
⑬ 取付穴

（φ６．
５ｍｍ）

⑭ 耳金付裏蓋⑦ ステム※耳金付裏蓋を使用する場合は、標準裏蓋のねじを全て外し、耳金付裏蓋と交換してねじで取り付けてください。

⑧ スピンドル⑨ 測定子②長針 …………… 目量を表す針です。

③長針用目盛板 … 目量を表す目盛です。⑩外枠を回すと回転します。

④止めねじ ……… ⑩外枠の回転を固定するねじです。
（WDI-0558SP・WDI-1058SP・WDI-0158SPのみ）⑤短針 …………… 目量×１００を表す針です。

⑥短針用目盛板 … 目量×１００を表す目盛です。

保持具のたわみ・浮きに注意する。

⑨測定子 ………… マスターやワークに当てる測定部です。
⑪標準裏蓋 ……… フラットなタイプの裏蓋です。
⑫耳金 …………… 保持具取付用パーツです。
⑭耳金付裏蓋 …… 耳金が付いた裏蓋です。

使用方法（準備）※保持例

トラブルシューティング■比較測定時、使用前と使用後で原点の位置がズレる①測定子と裏蓋にゆるみが無いことを確認する

・測定室の温度変化や、ダイヤルゲージの繰返し誤差が原因として考えられます。
下記の解決策をお試しください。
●測定室の温度が一定となるよう、環境を整える●測定時に任意の一定周期毎にマスタゲージを使用して、温度変化や繰返し誤差から来る原点ズレを調整する

ゆるみがある場合は、裏蓋の全てのねじを締め直してください。

②保持具にダイヤルゲージを取り付ける

ステムまたは耳金部で保持具に取り付けてください。
他の箇所での取り付けは、精度不具合や製品の破損の原因となります。

③長針・短針・スピンドルの動きが滑らかであることを確認する

指の腹で測定子をゆっくり押して上下させ、動きを確認してください。
滑らかでない場合は、ステムを締め付けすぎている恐れがありますので、調整してください。
また、針が設定位置で安定して静止することを確認してください。

④測定対象に対して、スピンドル（測定子）が垂直に接するよう設置する■測定値が安定しない、測定精度外の数値を示す

・測定子の磨耗が考えられます。
測定子が磨耗すると測定値に影響が出ます。定期的に磨耗確認を行い、測定値に影響が出る場合は交換を行ってください。

本体仕様

目

測定対象品量番

測定範囲

直角

目盛仕様
測定対象

使用方法（比較測定）

戻り誤差（mm）（mm）

WDI-10KD WDI-100.01(μm)

1/10回転指示誤差(μm)

8繰返し精密度(μm)
(g)

(mm)

① リミット針

質

量

測 定 子

交換用測定子品番保持位置を上下させるか、外枠を回して長針用目盛板を回転させ、長針を「０」に合わせます。

DI-1058 WDI-0342 WDI-0542 WDI-0558SP WDI-1058SP WDI-0158SP
0.01
0.01
0.001
351.5以下

全測定範囲指示誤差 (μm)
例）合格範囲：−０.０５〜＋０１. ５mm

10±0-100(N)

測 定 力

長針1回転②原点を設定する

1205±0-50DI-CP560.58145100-100-01.5以下

2以下

3981616100

鋼球

1±0-1001.5以下

180

鋼球

5131020190

20.2350.3

超硬球

DI-CPWDI-W

使用後のお手入れ・保管方法③マスターを外して、測定する

使用中に付着した埃、汚れを清掃する。※注油は行わないこと。

② 長針※リミット針について

リミット針は自由に動かして、合格範囲の目安として使うことができます。

使用方法（平行度・平面度・偏心測定等）①測定子を測定箇所に当てる

測定子を指の腹で軽くつまんで持ち上げて、上から降ろすように測定箇所に当てます。測定子の側面から衝突させないように注意してください。

・スピンドル摺動面の汚れは、乾いた布かアルコールを少量含ませた布で拭き取ってください。
・その他外観の汚れは、柔らかく乾いた布か、中性洗剤を少量含ませた布で拭き取ってください。

測定子の磨耗を確認する。

・測定子が磨耗すると測定値に影響が出ます。定期的に磨耗確認を行い、測定値に影響が出る場合は交換を行ってください。

付属のケースに収納し、乾燥した冷暗所に保管する。

・保管時、スピンドルに負荷をかけないよう、力（押し込み力・曲げ荷重など）をかけないでください。
・直射日光や湿気は避け、管理者以外が触れない状態で保管してください。

測定精度を保つため、定期校正を実施することをおすすめします。

保持位置を上下させるか、外枠を回して長針用目盛板を回転させ、長針を「０」に合わせます。

性能の劣化は、使用の頻度・環境によって大きく異なります。
お客様の社内規格に準じた周期で、定期点検を行うことをおすすめ致します。

測定対象をゆっくりと動かし、針の動きを読み取ってください。

当社にて校正を受け付けております。
全国にある当社代理店、取扱店を経由してお申し込みください。

②原点を設定する③測定対象を動かし、目盛を読み取る

なるべく短く取り付けの際、ステムを締め付けすぎない。
・ステムを締め付ける力が強すぎると、スピンドル動作不良の原因となります。

⑬取付穴 ………… 保持具取付部です。

スピンドルに衝撃を与えないようマスターを静かに外し、測定物を当てて目盛を読み取ってください。

支柱

保持具への取り付けは、ステムまたは耳金のみで行う。
・ステム、耳金以外での取り付けは、精度不具合や製品の破損の原因となります。

⑩外枠 …………… ③長針用目盛板を回転させます。

測定子を指の腹で軽くつまんで持ち上げて、上から降ろすようにマスターに当てます。測定子の側面から衝突させないように注意してください。

保持具

・保持具は十分に剛性のあるものを使用し、ダイヤルゲージの重みによる保持具のたわみを防いでください。
・保持具はダイヤルゲージの測定力に対して、十分に固定可能なものを使用し、測定時の保持具の浮きを防いでください。
・保持具の支柱はなるべく短く設定し、支柱のたわみを防いでください。
※たわみ、浮きは原点位置ズレや指示値のバラつきなど測定誤 ⑦ステム差の原因となります⑧スピンドル …… 測定時に上下する軸です。

①マスターを設置する

横方向へ力を加えない

ダイヤルゲージはマグネットベースやコンパレータスタンドなどの保持具で保持して使用します。
取り付けの際は下記の内容をお守りください。

⑦ステム ………… 保持具取付部です。

スピンドル（測定子）が垂直に接しないと、測定値や動作に不具合が出ます。
常に垂直を保つように設置してください。
※フライスのバイス平行確認に使用する際は、バイス測定面とスピンドル（測定子）の垂直を保ちつつ、フライス盤のテーブル送り時に支障とならないように、マグネットベースの位置を調整してください。

急激に動かさない

ご使用の前に⑥ 短針用目盛板①リミット針 …… 合否判定を素早く行うための、合格範囲の目印です。

スピンドルに衝撃を与えない。

・スピンドルを急激に動かしたり、横方向への力を加えないでください。精度不具合や故障、破損の原因となります。

下記の条件を満たす環境で使用する。

① リミット針（φ８ｍｍ）

・精度不具合や製品の破損の原因となります。
・止めねじを外すと、内部部品の歯車の噛み合いがズレて製品不具合の原因となる場合があります。

・指定用途以外へのご使用は、製品の破損や磨耗、予測できない事故の原因となります。

■各部の名称と機能 ※イラストはWDI-0558SPです。

③ 長針用
目盛板

分解、改造を行わない。

・取扱説明書に記載された内容以外での使用は、事故の原因となります。

●当商品に関するお問い合わせは、お買い求めの販売店もしくは当社にご連絡ください。

② 長針

必ず実行していただく内容『強制事項』です。

注意

安全上のお知らせ●第三者に譲渡・貸与される場合も、この説明書を必ず添付してください。

必ずお守りください。

発売元〒955-0055 新潟県三条市塚野目5丁目3番14号(0256)33-5502(代) FAX(0256)33-5528

URL http://www.niigataseiki.co.jp

E344-T22106