DIGITAL S-LINE INDICATOR（IP65）

INSTRUCTION MANUAL
Model No. DEI-233S1/DEI-233S2 <combined>

Thank you for purchasing the Digital S-Line Indicator.
For proper operation, please read this manual thoroughly before use.

PARTS IDENTIFICATIONFEATURES● IP65 class waterproof rating● Small size

Battery CoverLCD Display● ABS and REL measurement modes

ABS mode: With Display HoldREL mode: With one touch Preset recall● Direction of measurement (＋ / −) switchable● Preset function● Auto-off feature (approx. 4 hour)※REL mode onlyMODE[MODE] Button● Made in SwitzerlandSETSWISS MADE

[SET] Button

SPECIFICATIONS
DEI-233S1
DEI-233S20-12.5mm
0.01mm0.001mm0.01mm/0.001mm
0.01mm(switchable)

Model No.
Measurement Range

Accessories

Stem(Mounting Shaft)

Resolution

SpindleAccuracy

ＣＲ２０３２

CR2032(Lithium Cell)

Displayed Resolution

0.02mm
0.005mm
Less than 1.5N
0.002mmup to ±130mm
YES
φ8mm
5〜40℃
CR2032(Lithium cell)×170g

Measurement Force※Included for test

Contact Point

Repeatability
PresetMeasurement Dir. Switchable
Mounting Shaft Diameter
Operating Temp. Range

BATTERY REPLACEMENT①Remove the (2x) battery screws using (+) PH#0
screwdriver. Place a small standard screwdriver inthe notch to gently pry open the cover and slide outthe battery.
②Insert the new battery following the polarity markings
on the cover.
③Replace Battery Cover and tighten screws to secure.

Power Supply
WeightTROUBLESHOOTING
Error Condition

Corrective Action“Err 0” is Displayed
on LCD

Read Error.
Reset the sensor by pressing the[MODE] or [SET] button.
※If message is displayed again, there
is possibility that the memory iscorrupted; Please contact distributor
or place of purchase.

NotchMeasured value iswrong

Battery Cover●Battery CoverBattery
ＣＲ２０３２＋NOTICE Use only CR2032 type battery.

LCD does notdisplay● Displayed value isunstable● Display is hard toread●

Clean measuring surfaces with lenstissue or similar.
● Confirm and re-set 0-Point setting.
●

Cycle power by removing andreplacing battery.
● Replace battery with new one.

※If problem persists, or if you have any questions, please contact distributor
or place of purchase.
※Please note, manufacturer is unable to respond to inquires or provide
service directly. Please contact distributor or place of purchase.

See Reverse for “How to Use”→

FUNCTION SETUP
Press the [MODE] button to cycle through the functions inthe sequence shown below. When desired function isdisplayed,press [SET] button to access the various settings.
ABS[MODE] Button[MODE] Button【Measurement Screen】
Switchable between ABS or RELmodes.
＜ABS mode＞●Press [SET] Button to hold thedisplayed reading. Press onceand “H” icon will show, and value
will hold. Press again to release.
＜REL mode＞●At any probe position, press the[SET] button to call up anddisplay the Preset Value.
(see “Preset Value” below)【Preset Function】
Preset Value set can be calledup for display.
※For use in ABS, or REL mode.
【ABS / REL Mode】
To Set[MODE] Button

Press [MODE] to display ABS / REL,
then press [SET] to select.

※This function not available.
[MODE] Button【Preset Value】
Numerical value can be preset,
and later called up tomeasurement screen.
To Set[MODE] Button

Press [MODE] to move cursor, andpress [SET] to change value.

【Measurement Direction】
To Set[MODE] Button

Press [MODE] to display “Inc” (+) /“dEc” (-), then press [SET] to select.

【Resolution】※DEI-233S2 only
To Set[MODE] Button

Press [MODE] to display 0.01 / 0.001,
then press [SET] to select.

※If no operation is performed after 5 seconds, display
will return to the measurement screen.

POWER【Power ON】
Press [MODE], or [SET] button to turn on power.
【Power OFF】
Press and hold the [SET] button to turn off power.
※Power will switch off automatically if idle after
approximately 4 hours.
(※REL Mode only)※When battery level is low, the『 B 』icon is displayed;
please replace battery.

HOW TO USE①Mount indicator securely, such as on optional Magnetic
Base, and make sure that the Spindle is perpendicular
to the measurement surface.
②Press [MODE] or [SET] to turn on power.

③Setting 0-Point.
＜REL Mode＞
With gauge in contact with reference surface, press the[SET] button.
Display will read “0.00mm.”＜ABS Mode＞
With gauge in contact with reference surface, press the[MODE] button. When display reads “SEt”, press the[SET] button. (ref. “Function Setup - Preset Function”)
Display will read “0.00mm.”※If “0.00mm” is not displayed, please refer to “Function
Setup - Preset Value” to set value to 0.00mm, and thenperform steps ① to ③.
④With Gauge body held fixed, and the probe in contact
with the workpiece, the LCD will display the difference
between workpiece and reference plane.

CAUTION● This is a precision instrument, handle with care. Do notdrop or subject to excessive forces.

● Keep away form high voltage and strong magnetic fields;

instrument damage may result.

● Do not use or store in direct sunlight or locations withhigh temperatures such as in a car, or near a stove orheat source.
● When taking measurements, hold gauge securely, suchas in magnetic base. (sold separately)● Please do not disassemble or modify.
● Remove dust and cutting chips after use and apply rustpreventative oil to metallic surfaces. Please keep out ofreach of children.
● Do not use organic solvents to clean the body of theinstrument.
● When not being used for a long period of time, please
remove battery.
● Only use as directed. Improper use may cause accident
or injury.

Niigata seiki Co., Ltd.
6-15-22, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-31-5670 Fax. : +81-256-39-7730

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

I299-K1604