Flue Gas Analyzer HT-1300N
Operating Manual

１４１５
９．１
10. Ｇｕａｒａｎｔｅｅ

１５1

Reflecting plate

Connector for thermocouple(Option)*Ambient temperature isused to calculate e.g.
losses. In case ambient
temperature sensor is
not mounted, the fluegas temperature whichis measured during zerosetting will be assume
d as ambient airtemperature. And that
temp. will be fixed until
next zero setting.

During the measurement, press
During the pump off, press
During the pump on, press
During the pump off, press
Switch off the unit:
Back to the main menu.
Turn off the unit.

: Pump off ( Temp. measuremet will be continued): Pump in (Gas measurement starts again)

2times : Back to the main menu.
1time : Back to the main menu.

The functions "Input of T-Boiler" and "Input Soot number" has to be activatein the Extension Menu / Device settings / Parameter / Print / store boiler temp
and print / store soot number! (see chapter)８(with H2compensation)

0 ～ 650℃
0 ～ 1100℃ (depend on probes)±1℃ or ±1％ reading(0 ～ 650℃）±２％ reading (650.1 ～ 1100℃）(0 ～ 999.9℃） 1℃（1000 ～ 1100℃）(with H2compensation)(AC100-240V 50/60Hz DC12V 2700mA)

Standardequipment

Built inPump, Interface for PC (RS232)
Data logger (possible to store 100 data), interface for printerAccessories
Battery chager, gas sampling probe, ambient air temp. sensor, reflecting plate, carrying case, softcase
Description
Art. No.
L=180ｍｍ φ5 0～650℃*Sampling probe with K thermocouple ＨＴ-1006

Optional items

Probes for flue gas
HT-1001A
HT-1006L
Probe handle
HT-7201A
Probe handle
Probe tube（for probe handle HT-7201G）
Probe tube（with flue gase temp. sensor） HT-7231
HT-7232
HT-7233
K thermocouple HT-1251a
Probes fortemperature
HT-1252a
HT-1253a
HT-1254a
HT-1255aHT-1256a
HT-1257a
Air temp. sensor
HT-1382
Ambient air temp. sensor
HT-1321
Battery charger
HT-1318
Hard caseHT-1315
Measurement software
HT-2094
Measurement software
HT-2084
Infrared printerHT-1610Roll paper for infrared printerHT-1636*In case measure sulfur-rich flue gas, acctual dew point might differ from calculated dew point.

* Fuel data might slightly differ depend on location,
therefor calculated data also might have difference.

Sampling probe
Ｌ-shape probe

L=300mm φ6
L=120mm φ60～650℃0～650℃

with hose/condensate trap
Ｌ=180ｍｍ φ5
Ｌ=750ｍｍ φ6
Ｌ=750ｍｍ φ8
φ3×130L, 0～950℃, for air /liquid
φ1.5×130L, 0～950℃, for air /liquid
φ3×130L, 0～400℃, for air /liquid/foods,centric top
130L, 0～400℃, for surface/rifts/air /liquid
φ4×130L, 0～650℃, for surface/rifts/air /liquid0～450℃, magnetic probe for surface0～180℃, Pliers probe for plates, tube0～100℃、for air temp.
0～100℃
ＡＣ100～240V
Aluminum, dimentino：350×460×155 weight：2.9ｋｇ
Online View 2000（OS Windows XP/7） with RS232cable
Online View 2000（OS Windows XP/7）with RS232cable,USBcablewith roll paper×1 , AAbattery x 45rolls

9910. Guarantee
Guarantee period : 12months from date of dispatch.
Guarantee: During guarantee period, if your instrument brakes down although correct usage basedon this user manual, we will repair it by free of charge. In case you have trouble,
firstly contact to HODAKA CO., LTD. (TEL +81-(0)6-6922-5501 / E-mail: info@hodaka-inc.co.jp )then send your instrument to HODAKA CO., LTD.
International transportation cost is not include in guarantee.
HODAKA CO., LTD shall not be liable for any loss or damage whatever arising from content errors
or any misuse of this instrument.
HODAKA CO., LTD.
1-6-17 Asahiku Takadono Osaka Japan 535-0031
TEL: +81-(0)6-6922-5501
FAX: +81-(0)6-6923-1617
Traceability certification can be issued at HODAKA. ( Additional cost will be required)

HODAKA CO,. LTDver.140201

1-6-17 Takadono, Asahi-ku, Osaka 533-31 Japantel +81-6-6922-5501 fax +81-6-6922-5895
e ‐ mail ； info@hodaka-inc.co.jp
UPL ； http://www.hodaka-inc.co.jp