User Manual
PRECISION MEASURING TOOLS

Model No. GDCS-＊＊＊DIGITAL CALIPERS

Thank you for purchasing Niigata Seiki Digital Calipers.
This is a precision measuring tool for making Inside, Outside, Step Height, and Depth measurements.

●For safe and proper use, please read this instruction manual before use and follow the procedures described.
Please keep manual where it is accessible to user for future reference.
●Keep this manual with the instrument if transferred or leased to a third party.
●For inquiries about this product, please contact dealer or Niigata Seiki.

■PART IDENTIFICATION AND FUNCTION⑤ Measuring surfaces forinside measurements⑩LCD Display⑥Depth Bar⑧Scale①Measuring surfaces foroutside measurements※Models GDCS-150W, GDCS-200W,
GDCS-300W have carbide tipped Inside
and Outside Jaws.

※Model GDCS-150CD
has is φ1.9mm
round Depth Bar.

⑪ Thumb Wheel⑫ Battery Cover⑬ [ZERO] Button⑭ [ON/OFF] Button⑦Measuring surfaces
for depth measurements

To prevent harm to yourself or others, and to prevent damage toproperty, always follow the procedures marked by the followingsymbols.

Denotes a requirement You MUST doIndicates risk of personal injury or property damage if not followed.

Read the manual and follow the directions.
・Use of product other than as described in this manual
may cause accident.
Use only for measuring.
・Use for any purpose other than measuring may
damage or wear the instrument. Improper use may
also cause accident.
Handle with Care.
・Do not drop or subject instrument to excessive shock.
Do not place under heavy objects. Improper handling
may cause damage or poor accuracy.
・Do not scratch instrument, for example by writing ID
number.

●Use care to avoid damaging
terminals when inserting
battery.

Use and store only in environment meeting thefollowing conditions:●Dry location protected from rain, water, and oils.
●Location that is not excessively hot.
●Location protected from use by children andunauthorized people.
・Use in places contrary to the above may cause
damage to the product resulting in accident or injury.
Jaw Tips Are Sharp̶Handle With Care.
・Careless handling may cause injury.
Do not disassemble or modify.
・Disassembly may damage product or cause accident.
For repair, please contact distributor or place ofpurchase.
Do not write on instrument with electric engraver.
・Do not engrave instrument body or jaw, such as an ID
number with engraver or electric pen.

■Power ON/OFF①To turn on, press the [ON/OFF ] Button.
②To turn off, press the [ON/OFF ] Button.
③With power on, press the [HOLD] Button to fix the value on
the display.

■Absolute Measurement Mode

①Loosen the lock screw, and turn on power.
②With jaws closed, press the [ZERO] Button.
③When “0.00mm” is displayed on LCD, Calipers is ready tostart measurement.

■Comparisons Measure Mode

NoticeBattery
CoverBattery

Terminals

HOW TO USE①Loosen the lock screw, and turn on power.
②With jaws open to desired reference position (any point isOK,) press [ZERO] Button.
③When “0.00mm” is displayed on LCD, Calipers is ready tostart measurement.

●Terminals may be broken if battery is pressed straight down.
●Insert battery at angle, underneath the small hook, as shown.

Please Observe
Denotes a prohibition You MUST NOT doUse only CR2032 typebattery.

Pressed Down

SAFETY PRECAUTIONS

Avoid using in high electric fields, such as nearfluorescent lights or switching power supplies.
・Incorrect value may be displayed on LCD.

① Slide cover in direction of arrow.
② Insert battery with (+) side outward.
③ Replace battery cover.
Notice①Measuring Surfaces for outside measurements … Object to be measured is positioned between these surfaces.
②Outside Jaws ……………………………………… For measuring outside dimensions.
③Measuring surfaces for step measurements …… Measures height of steps.
④Inside Jaws………………………………………… For measuring inside dimensions.
⑤Measuring surfaces for inside measurements …… Position surfaces inside object to be measured.
⑥Depth Bar ………………………………………… For measuring depth.
⑦Measuring surfaces for depth measurements …… Depth is measured between these two surfaces.
⑧Scale ……………………………………………… Supports LCD Display Unit.
⑨Lock Screw ……………………………………… Tighten to Prevent the Slide and Display from moving.
⑩LCD Display ……………………………………… Displays digital measurement.
⑪Thumb Wheel …………………………………… For adjusting position of Display Slide.
⑫Battery Cover …………………………………… Open to access battery.
⑬[ZERO] Button …………………………………… When pressed, Display will read "0.00".
⑭[ON/OFF] Button ………………………………… For turning power ON/OFF.
⑮[HOLD] Button …………………………………… When pressed, "HOLD"Icon appears on Display and measurement will be fixed.
⑯Output Connector ………………………………… Measurement value can be outputted to the PC by connecting BluetoothBOXof the option.

WARNING

Contamination of surfaces may cause measurement error.

●Allow the calipers and object to be measured time toreach the same temperature.

vemoRe③ Measuring surfaces forstep measurements②Outside Jaws

●Wipe off rust proofing and dirt from measuring
surfaces and slide.

INSERTING・REPLACING BATTERY⑮ [HOLD] Button④ Inside Jaws

Before using the calipers, please follow the steps below:

A temperature difference between the object to be measured
and the caliper may cause measurement error. Allow enoughtime for the temperatures to equilibrate.

⑨ Lock Screw⑯ Output Connector

PREPARATION FOR USE

HookIf Jaws are opened beyond maximum position,Thumb Wheel may come loose. Please use care.

AFTER USE CARE・STORAGE●After use, wipe off any dirt or oil from measuring
surfaces, sliding surfaces, and exterior using a dry
cloth and perform rust prevention treatment.
For rust prevention, lightly apply anti-rust oil, or place in rustpreventive bag.

●Store with Jaws slightly open.

Slide InIf Jaws are closed, expansion due to temperature changes may
cause excessive force which will cause inaccuracy.

Terminals
Battery●Store in a cool, dry, and dark location in the providedcase.
Keep out of direct sunlight and moisture, and please keep
secure from unauthorized personnel.

TROUBLESHOOTING
Q. Nothing is displayed on LCD.
A. The battery may be running low, please replace battery
with a new one.
Q. With power on, the number on the LCD does not change
as Display Unit is moved.
A. Please reset by removing battery for about 30sec., andthen replacing.
Q. Nothing happens when buttons are pressed.
A. Make sure battery is properly inserted.
Q. The characters “MIN”, “MAX”, or “△” are displayed on LCD.
A. Cycle power, or apply several long presses to [HOLD]
Button. Use as normal once character is off.
Q. The correct value is not displayed.
A. High humidity may cause dampness. Please dry off the
instrument and use only in dry location.

SERVICE●If Instrument is not working properly, or if you have any
questions, please contact distributor or place of purchase.
●Please note, manufacturer is unable to respond to inquires orprovide service directly. Please contact distributor or place ofpurchase.

●Remove battery if Caliper will not be used for a long
period of time.

PRODUCT SPECIFICATIONS
Measuring Range
Model No.
0 -100mm
GDCS-100
0 -150mm
GDCS-150
0 -200mm
GDCS-200
0 -300mm
GDCS-300
0 -150mm
GDCS-150CD
0 -150mm
GDCS-150W
0 -200mm
GDCS-200W
0 -300mm
GDCS-300W

Accuracy※±0.03mm±0.04mm±0.03mm±0.04mm

Weight155g
175g
200g
260g
175g
175g
200g
260g

※Not including quantization error (±1 count)●Resolution : 0.01mm●Display: 5 Digit LCD●Auto Off: approx. 20 min.
●Maximum response speed : Slide to Right ..... 1.5 m/s
Slide to Left ....... 0.75 m/s※Left/Right as shown in diagram at top of manual.

●Operating Temperature Range :0°
C ~ 40°C●Storage Temperature Range : −10°
C ~ 60°C●Power: (1x) CR2032 Lithium Battery●Battery Life : approx. 2.5 years in normal use●Accessory : CR2032 Lithium Battery (for test)

Niigata seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

取扱説明書 品番 GDCS-****

精密測定工具デジタルノギス●測定面、摺動面、外観などの汚れや防錆油などを拭き取る
そのまま測定を行うと誤差の原因となります。

●本体と被測定物の温度慣らしを十分に行う●正しく安全にお使いいただくため、ご使用の前に本取扱説明書を必ず読み、記載の手順に従ってご使用ください。
お読みになった後は、お使いになる方がいつでも見ることができる場所に大切に保管してください。
●第三者に譲渡・貸与される場合も、この説明書を必ず添付してください。
●当商品に関するお問い合わせは、お買い求めの販売店もしくは当社にご連絡ください。

⑨ 止めねじ

⑯ 出力コネクタ④内側用ジョウ

⑧本尺⑥デプスバー※GDCS-150CD のデプス バーは丸棒（φ1.9 ｍｍ）タイプです。

③段差測定面②外側用ジョウ

⑦ 深さ測定面⑪ サムローラ⑫電 池蓋⑬ZEROスイッチ①外側用測定面⑭ ON/OFFスイッチ※GDCS-150W、GDCS-200W、GDCS-300W は、外側、内側用測定面に超硬合金チップが付いています。

⑨止めねじ …………… デジタル液晶部のスライドを固定します。

①外側用測定面 …… 被測定物をこの面で挟んで測定します。

⑪サムローラ ………… デジタル液晶部をスライドさせます。

③段差測定面 ……… 段差面に当てて長さを測定します。

⑬ZEROスイッチ …… 押すと画面数値が"0.00"と表示されます。

④内側用ジョウ …… 被測定物の内側を測ります。

⑭ON/OFFスイッチ … 電源のON/OFFを行います。

⑤内側用測定面 …… 被測定物の内側にこの面を当てて測定します。 ⑮HOLDスイッチ …… 押すと画面上に"HOLD"と表示され、表示している数値を固定できます。
⑥デプスバー ……… 深さを測定するバーです。
⑯出力コネクタ ……… 別売の Bluetooth BOX を接続すること⑦深さ測定面 ……… 測定面に当てて深さを測定します。
で、測定値をパソコンへ出力すること⑧本尺 ……………… デジタル液晶部のスライドを保持します。
ができます。

注意

必ずお守りください

測定作業にのみ使用する。
・指定用途以外へのご使用は、製品の破損や磨耗、予測できない事故の原因となります。
大切に取り扱う。
・ぶつける、落とすなどの衝撃を与えたり、重量物を上に乗せたりしないでください。精度不良や破損の原因となります。
・商品に傷をつけないでください。
インバータ式蛍光灯などの電界が発生する場所での使用は避ける。
・ＬＣＤ表示部の数値が誤作動を起こす原因となります。

電池は必ずCR2032をお使いください。

電池蓋してはいけない内容『禁止事項』です。

必ず実行していただく
内容『強制事項』です。

下記の条件を満たす環境で使用・保管する。
●雨や水、油などがかからない、乾燥した場所●高温にならない場所●子どもや、使用者以外が近付かない場所
・上記に反する場所での使用は製品の破損、事故やけがの原因となります。
ジョウ先端は尖っているので、扱いに注意する。
・けがの原因となります。
分解・改造を行わない。
・製品の破損や事故の原因となります。修理はお買い上げの販売店、または発売元までご連絡ください。
電気ペンなどで書き込まない。
・電気ペンなどで本尺やジョウなどに、番号などの書き込みをしないでください。

■絶対測定モード①止めねじをゆるめ、電源を入れてください。
②ノギスのジョウを閉じた状態で "ZERO"スイッチを押してください。
③LCD表示部に"0.00mm"が表示されると計測開始ＯＫです。

ご注意●電池をセットする際は、端子の折れにご注意ください。

●電池をセットする際に上から強く押し込むと、端子が折れて破損する場合があります。
●電池をセットする際は、右斜め方向から端子を外側に押すよ
うに、本体のツメの下へ入れ込んでください。
電池

端子①電源をONにするには、”ON/OFF”スイッチを押してください。
②電源をOFFにするには、”ON/OFF”スイッチを押してください。
③電源ONの状態から“HOLD”スイッチを押すと数値が固定されます。

①止めねじをゆるめ、電源を入れてください。
②ノギスのジョウが開いた状態（どのポイントでもＯＫ）の時に"ZERO"スイッチを押してください。
③LCD表示部に"0.00mm"が表示されると計測開始ＯＫです。
ジョウを最大まで開くと、サムローラが外れることがありますので、ご注意ください。

使用後のお手入れ・保管方法●測定面・摺動面・外観などの汚れや油などを乾いたウエスで拭き取り、防錆処理を行う

防錆処理は防錆油を薄く塗るか、防錆袋などに入れてください。

●ジョウを少し開いた状態で保管する

ジョウを閉じた状態で保管すると、温度変化による膨張などで無理な力がかかり、精度不良の原因となります。

ツメ
入れ込む

端子

●付属のケースに入れ、乾燥した冷暗所に保管する電池

直射日光や湿気は避け、管理者以外が触れない状態で保管してください。

●長期間使用しない場合は電池を製品から取り外した状態で保管してください

トラブルと対応

製品仕様

Ｑ．LCD表示部に何も表示されない。
Ａ．電池の残量が少なくなっている可能性があります。新しい電池と交換してください。

品番
GDCS - 100
GDCS - 150
GDCS - 200
GDCS - 300
GDCS - 150CD
GDCS - 150W
GDCS - 200W
GDCS - 300W

Ｑ．電源ONの状態でスライドさせてもLCD表示の数字が変わらない。
Ａ．一度電池を取り外し、30秒後に電池をセットし直してください。

お守りいただかないと『傷害を負う、または物的損害が発生するおそれがある内容』です。

取扱説明書をよく読み、指示に従う。
・取扱説明書に記載された内容以外での使用は、事故の原因となります。

ご注意■電源のON/OFF■比較測定モード①電池蓋を矢印の向きに開けます。
②電池を、必ず（＋）側を表にしてセットしてください。
③電池蓋を元通りに装着してください。

押し込む⑫電池蓋 ……………… 開けて電池をセットします。

お使いになる人や他の人への危害、財産への損害を未然に防止するた
め、必ずお守りいただくことを右記のように説明しています。

電池のセット・交換⑩LCD表示部 ………… 測定数値をデジタル表示します。

②外側用ジョウ …… 被測定物の外側を測ります。

安全上のご注意

測定器本体と被測定物の間に温度差があると、誤差の原因となります。同じ温度になるよう、温度慣らしは十分に行ってください。

す外⑤内側用測定面⑩LCD表示部⑮ HOLDスイッチ

使用方法

ご使用前に、下記の準備を行ってください。

この度は「デジタルノギス」をお買上げいただきありがとうございます。
この商品は、測定物の外側・内側・段差・深さ測定が可能な精密測定工具です。

■各部の名称と機能使用前の準備

Ｑ．スイッチを押しても何も反応しない。
Ａ．電池が正しくセットされているか確認してください。

Ｑ．LCD表示部に『MAX』『MIN』『△』などの文字が表示されている。
Ａ．一度電源を切るか、HOLDスイッチを何回か長押しして文字を消してからご使用ください。

Ｑ．正しい数値が表示されない。
Ａ．使用環境の湿度が高い可能性があります。本器を乾燥させ、湿度が低い環境下でご使用ください。

修理の際は●正常に作動しない場合や、不明な点がございましたらお買い上げの販売店または、発売元までご連絡ください。
●お問い合わせや、ご連絡が無いまま直接発売元に修理品などを送付されても処理、対応ができない場合がありますのでご了承ください。

※量子化誤差（±1カウント）を含まず測定範囲（mm）※器 差（mm） 質 量（g）
0 - 100155
0 -150±0.03
175
0 -200200±0.04
0 -300260
175
0 -150±0.03
175
0 -150200
0 -200±0.04
260
0 - 300●最 小 読 取 値：0.01mm
示：LCD 5桁表示
●表
●オ ー ト オ フ：約20分●最大応答速度：右方向へのスライド…1.5m/s
左方向へのスライド…0.75m/s※右方向、左方向は各部の名称の図を正面にした場合●使用温度範囲：0℃〜40℃●保存温度範囲：−10℃〜60℃
源：リチウム電池 CR2032×1●電●電 池 寿 命：通常使用状態で約2.5年
属品：リチウム電池 CR2032（テスト用）●付発売元〒955-0055 新潟県三条市塚野目5丁目3番14号(0256)33-5502(代) FAX(0256)33-5528

URL http://www.niigataseiki.co.jp

20111000

E1102-Q1