DIGITAL HEIGHT GAUGEINSTRUCTION MANUAL
Model No.:VH-30D / VH-60D <combined>

Thank you for purchasing the Digital Height Gauge.
Please read this manual thoroughly before use for proper operation.

PART IDENTIFICATIONAPPLICATIONS and FEATURES【ACCESSORIES】● Ideal for measuring and marking work for molds, jigs, and tooling.
● Accessory arm for mounting φ8mm or φ6mm dial indicator.
● Easy to ready digital display.
● Origin can be set at any position. ● Preset function.
● TOL function for convenient inspection work.
● HOLD function.

PostIndicator Mounting ArmIndicator Clamp Screw

VH-30D
φ8mm
Mounting HoleSPECIFICATIONS(with φ6mm collet)

VH-30D

VH-60D

Measurement Range
0-300mm0-600mm
Resolution
0.01mmMaximum permissible error ※±0.04mm±0.06mmFine Adjustment Range
approx. 8mm
Display
5 digit LCD
Weightapprox. 2.6kg approx.5.0kg
Operating Temperature Range
0~40ºC

Fine Adjust Lock Screw
Fine Adjust Carriage[ON/OFF/ZERO] Button
Carriage Lock Screw

LCD Display[HOLD] Button[TOL] Button

Scribe Clamp Screw
Jaw・Before using for the first time, please wipe off the corrosion
protectant cloth.
・This is a precision instrument, handle with care. Do not dropor subject to excessive forces.
・Please avoid damage to post. Also, do not scratch instrument,
for example by writing ID number.
・Keep away from direct sunlight. Do not store instrument inextreme temperatures.
・When measuring using a dial indicator in place of scribe,
listed values for Maximum permissible error may not beachieved.

Base(w/ Carbide Tip)

Scribe Clamp
Base Reference Plane

PREPARATIONS BEFORE USE
1 Inserting Battery

3 Indicator Mounting Arm

1. Open Battery Cover by sliding in direction of arrow (2. Insert battery with positive (+) side facing out.
3. Replace Battery Cover.
Caution: Only use SR44 Silver Oxide Battery.

).

※Remove battery if Instrument will not be used for along period of time. Batteries may leak if left inunused instrument for an extended period.

2 Installing the Scribe
1. Loosen the Scribe Clamp Screw and insert onto Jaw.
(Figure : ①)
2. Insert the Scribe to desired overhang and secure bytightening the Scribe Clamp Screw. (Figure : ②)

The Indicator Mounting Arm is used to mount a dial indicator (sold separately)in place of the Scribe.
1. Loosen the Indicator Clamp Screw, insert the indicator shaft, andsecure by tightening the clamp screw.
※ For indicators with a φ6mm mounting shaft, please attach using
φ6mm collet.
2. Loosen the Scribe Clamp Screw and clamp onto the Jaw.
3. Insert the Indicator Mounting Arm into the clamp, and tighten Clamp
Screw at the desired amount of overhang.

Jaw

Dial Indicator(sold separately)

Indicator Clamp Screw

ScribeIndicator Mounting Arm

Scribe Clamp Screw
Scribe②Scribe Clamp

JawVH-60D
Dial Indicator(sold separately)

ScribeClamp Screw
Indicator
Mounting Arm

2 Comparative (Relative) Measurement
1. Press the [ON/OFF/ZERO] Button to turn on the power.
2. Measure the reference surface and press the [ON/OFF/ZERO]
Button to set the LCD display to “0.00mm”
3. Measure the surface to be compared, and the measurement
relative to the reference surface will be displayed.

Scribe(Figure:1)

Measuring Surface

Surface Plate

Reference Plane

Fine Adjust Lock Screw
Fine Adjust Carriage

１ Tighten

Carriage Lock Screw

２ Loosen

1. Tighten the Fine Adjust Lock Screw, and loosen the Carriage
Lock Screw. (Figure:2)
2. Turn the Fine Adjust Carriage to the desired height setting.
(Figure:2)

BUTTON OPERATIONTOL Button (Setting tolerance range)

1 ON/OFF/ZERO Button

4● Press when power is off ....................... Power ON● Press when power is on ....................... Display value reset● Press and hold when power is on ........ Power OFF

The [TOL] Button is used to confirm that a measured value is
within a specified range of tolerance.

2 HOLD Button (Freeze display)Press the [HOLD] Button to freeze the display value. “H” icon will
display on LCD to indicate hold.

3 SET Button (Preset value)[SET] Button allows you to preset a value.
1. When [SET] Button is pressed, “SET“ icon will blink on LCD.
2. Long press of the [SET] Button selects which character tomodify. Release the button when the desired character isblinking, and a short press modifies the number at that
position. When the “SET” icon is blinking, press [SET] again to
store the value.
The preset value is now entered.
※ When [ON/OFF/ZERO] Button is pressed, the display will beset to the preset value.
To return the preset to “0.00mm,” enter that value using theabove procedure, or toggle power by removing andreplacing the battery.

1. When you press the [TOL] Button, “ “ is displayed on theLCD, and “SET” icon will blink.
In this state the maximum value for the tolerance range is set.
2. Long press of the [TOL] Button selects which character will
blink, and a short press modifies the value at that position.
Continue to enter the desired value and return the blinking tothe “SET” icon. Press [TOL] again to continue.
3. “ ” will display on LCD; repeat the procedure from step 2 toset the minimum value for the tolerance range.
4. When complete and the “SET” icon is blinking, press [TOL].
Tolerance range is now set.
5. While taking measurements in tolerance mode,● If the measurement is within the tolerance range, “OK” isdisplayed on the LCD.
● If the measurement is above the tolerance range, “ “ isdisplayed on the LCD.
● If the measurement is below the tolerance range, “ “ isdisplayed on the LCD.
※In tolerance mode, the [ON/OFF/ZERO] Button does notfunction. To use the [ON/OFF/ZERO] Button, press [TOL] tocancel tolerance mode.

【LCD Display】

Q. Nothing is displayed, or the numbers are flashing.
A. Is the battery running low?
Q. With power on, the number on the LCD does not change as
the carriage is moved.
A. Please reset by removing battery for about 30sec., and thenreplacing.
Q. Nothing happens when buttons are pressed.
A. Is the battery inserted properly?

SERVICE● If height gauge is not working properly, or if you have anyquestions, please contact distributor or place of purchase.

Jaw
Scribe Clamp

Base

3 Fine Adjust Feed Operation● Please note, manufacturer is unable to respond to inquires or
Overhang Adjustment

LCD Display

1. Press the [ON/OFF/ZERO] Button to turn on the power.
2. Bring the scribe down to the surface plate (the base referenceplane) and make good contact. (Figure:1)
3. Press the [ON/OFF/ZERO] Button to set the LCD display to“0.00mm”
4. Position the scribe to the workpiece to take a measurement.

TROUBLESHOOTING

ScribeClamp Screw[ON/OFF/ZERO]Button

1 Absolute Measurement

VH-30D①Scribe Clamp ScrewScribe Clamp※Not including quantization error (±1count)

NOTES

Battery Cover

Measuring Surface

・Indicator Mounting Arm ・Scribe
・Scribe Clamp ・φ6mm Collet
・SR44 Silver Oxide Battery (for test)

Accessories[SET] Button

Scribe

VH-30DMEASURING

provide service directly. Please contact distributor or place ofpurchase.

Niigata seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

デジタルハイトゲージ

取扱説明書

品番：ＶＨ-３０Ｄ／ＶＨ-６０Ｄ ＜兼用＞

この度はデジタルハイトゲージをお買上げ頂きありがとうございます。
ご使用前に、本書をよくお読み頂き正しくお使いくださいますようお願い申し上げます。

各部の名称

用途・特長

【付属品】
ホルダアームホルダ止めねじ●金型・治工具の測定やケガキ作業に最適●付属品のホルダアーム使用でφ６・８mm 軸のダイヤルインジケータを取り付けることができます。
●読み取りやすいデジタル表示。
●任意の位置で０設定ができます。 ●検査に便利なＴＯＬ機能付。
●数値のプリセット機能付。
●ホールド機能付。

柱

VH-30D用

φ８用取付穴仕（φ６コレット付）

VH-60D用測 定 範 囲最 小 読 取 値
最 大 許 容 誤 差※
スライダ微動送り上下量表示
質量
使 用 温 度 範 囲

微動止めねじ
微動送り車
微動送りねじＬＣＤ表示部ＨＯＬＤボタンスクライバ止めねじ

ＯＮ／ＯＦＦ／ＺＥＲＯ
スイッチ
操作パネル止めねじ
ＴＯＬボタンスクライバベース

測定面（超硬チップ付）スクライバクランプ
ベース基準面

使用前の準備

１ 電池の入れ方

１．電池蓋を の向きにスライドして開けます。
２．電池は必ず（＋）側を表にしてセットしてください。
３．電池蓋をもとの位置に戻してください。

ご注意 電池は必ず酸化銀電池ＳＲ４４をお使いください。
※長期間使用しない場合は電池を製品から取り外した状態で保管してください。長期間使用しないで放置すると、電池の液漏れの原因となります。

２ スクライバの取り付け

１．スクライバクランプのスクライバ止めねじをゆるめ、ジョウに通します。
（①）
２．スクライバをスクライバクランプに入れ、突き出し調節を行った後、スクライバ止めねじを締めて固定してください。
（②）①スクライバ止めねじ

スクライバクランプ

属品

ＶＨ-３０Ｄ
０-３００mm
±０．
０４mm

ジョウ

※量子化誤差（±１カウント）含まず

・はじめてのご使用前には、柔らかい布などで防錆油を拭き取ってください。
・本品は精密測定機器です。落下等の強いショックを加えないでください。
・電気ペンなどでの番号の書き込みはさけてください。また、柱にキズをつけないでください。
・直射日光下での使用や極限温度下での保管はさけてください。
・ダイヤルインジケータを使用して測定する際には、スクライバよりも測定点が前に出るため、記載されている最大許容誤差に満たない場合があります。

３ ホルダアームについてスクライバの替わりに付属のホルダアームを使用して、別売のダイヤルインジケータを取り付けることができます。
１．ホルダアームのホルダ止めねじをゆるめ、取付穴にダイヤルインジケータの軸を差し込んで、止めねじを締めて固定してください。
※ダイヤルインジケータの取付軸がφ６ｍｍの場合は、φ６コレットを使用して取り付けてください。
２．スクライバクランプのスクライバ止めねじをゆるめ、ジョウに通します。
３．ホルダアームをスクライバクランプに入れ、突き出し調節を行った後、スクライバ止めねじを締めて固定してください。
VH-30D

スクライバ

突き出し調節スクライバクランプ

VH-60D
ダイヤルインジケータ
（別売品）

スクライバ
クランプ

ＯＮ／ＯＦＦ／ＺＥＲＯ
スイッチ

１ 絶対測定ＬＣＤ表示部

１．ＯＮ／ＯＦＦ／ＺＥＲＯスイッチを押して、電源を入れます。
２．スクライバの測定面を定盤（ベースの基準面）に密着させます。
（図１）
３．ＯＮ／ＯＦＦ／ＺＥＲＯスイッチを押して、ＬＣＤ表示部の数値を「０.００ｍｍ」にしてください。
４．被測定物にスクライバを当てて測定してください。

ベーススクライバ

２ 比較測定（図１）

１．ＯＮ／ＯＦＦ／ＺＥＲＯスイッチを押して、電源を入れます。
２．基準となる被測定物を測定し、ＯＮ／ＯＦＦ／ＺＥＲＯスイッチを押して、ＬＣＤ表示部の数値を「０.００ｍｍ」にしてください。
３．比較する被測定物を測定してください。基準となる被測定物との差が表示されます。

測定面

定盤

１ 締める

微動送り車

３ 微動送り機能

操作パネル止めねじ

１．微動止めねじを締め、操作パネル止めねじをゆるめます。
（図２）
２．微動送り車を回し、数値合わせを行ってください。

２ ゆるめる（図２）

各ボタンの使用方法
１ ＯＮ／ＯＦＦ／ＺＥＲＯスイッチ●電源ＯＦＦの状態で１回押す … 電源ＯＮ
●電源ＯＮの状態で１回押す…… 表示数値のリセット●電源ＯＮの状態で長押し……… 電源ＯＦＦ

２ ＨＯＬＤボタン（数値の固定）

ＨＯＬＤボタンを押すと、ＬＣＤ表示部に「Ｈ」が表示され、数値が固定されます。

３ ＳＥＴボタン（数値のプリセット）

ＳＥＴボタンの使用で、任意の数値のプリセットができます。
１．ＳＥＴボタンを押すと、ＬＣＤ表示部の「ＳＥＴ」が点滅します。
２．ＳＥＴボタンを長押しすると点滅位置が変わります。
点滅している場所で一度ＳＥＴボタンを放し、再度押すと＋−、または数字が切り替わります。
これを繰り返し、任意の数値に設定したら点滅を「ＳＥＴ」に戻し、ＳＥＴボタンを押してください。
これで数値がプリセットされました。
※この状態で測定中にＯＮ／ＯＦＦ／ＺＥＲＯスイッチを押すと、ＬＣＤ表示はプリセットした数値にリセットされます。
リセットした数値を「０.００ｍｍ」に戻したい場合は、一度電池を抜くか、上記の手順でプリセット数値を「０.００ｍｍ」に合わせてください。

基準面

微動止めねじ

４ ＴＯＬボタン（公差の設定）

ＴＯＬボタンの使用で、測定値が指定した公差の範囲に入っているかを簡単に確認することができます。
１．ＴＯＬボタンを押すと、ＬＣＤ表示部の「ＳＥＴ」が点滅し、「 」が表示されます。
この状態で公差の最大値を設定します。
２．ＴＯＬボタンを長押しすると点滅位置が変わります。
点滅している場所で一度ＴＯＬボタンを放し、再度押すと＋−、または数字が切り替わります。
これを繰り返し、任意の数値に設定したら点滅を「ＳＥＴ」に戻し、ＴＯＬボタンを押してください。
３．「 」が表示されますので、公差の最小値を２．と同様の手順で、設定してください。
４．点滅を「ＳＥＴ」に戻し、ＴＯＬボタンを押してください。
これで公差の範囲が設定されました。
５．測定を行います。
● 測定値が公差の範囲に収まっている場合、ＬＣＤ表示部に「ＯＫ」が表示されます。
● 測定値が公差の範囲を上回っている場合、ＬＣＤ表示部に「 」が表示されます。
● 測定値が公差の範囲を下回っている場合、ＬＣＤ表示部に「 」が表示されます。
※ＴＯＬ設定状態では、ＯＮ／ＯＦＦ／ＺＥＲＯスイッチは機能しません。ＯＮ／ＯＦＦ／ＺＥＲＯスイッチを使用する場合は、再度ＴＯＬボタンを押し、ＴＯＬ設定を解除してください。

トラブルと対応

スクライバ
止めねじ

ダイヤルインジケータ（別売品）

ホルダアーム②±０．
０６mm

・ダイヤルインジケータ取付用ホルダアーム・スクライバ ・スクライバクランプ
・φ６コレット ・酸化銀電池ＳＲ４４（テスト用）

ホルダ止めねじ

スクライバ止めねじ

０．
０１mm

ＶＨ-６０Ｄ
０-６００mm

約８mm
ＬＣＤ５桁表示
約２.６kg約５.０kg０〜４０℃

ご使用にあたって

電池蓋

スクライバ付ＳＥＴボタン

ジョウ

様

測定方法

【ＬＣＤ表示部】

Ｑ．画面に何も表示されなかったり、数字が点滅している。
Ａ．電池の残量が少なくなっていませんか？
ジョウ

スクライバ
止めねじ

ホルダアームジョウ

Ｑ．電源ＯＮの状態でスライドさせてもＬＣＤ表示の数字が変わらない。
Ａ．一度電池を取り外し、３０秒後に電池をセットし直してください。
Ｑ ．スイッチを押しても何も反応しない。
Ａ．電池が正しくセットされていますか？

修理の際は●正常に作動しない場合や、不明な点がございましたらお買い上げの販売店または、発売元までご連絡ください。
●お問い合わせや、ご連絡が無いまま直接発売元に修理品などを送付されても処理、対応ができない場合がありますのでご了承ください。

発売元〒955-0055 新潟県三条市塚野目5丁目3番14号(0256)33-5502(代) FAX(0256)33-5528

URL http://www.niigataseiki.co.jp
2207
E1161-Q