Flue gas analyzer

HT-1300N

15O2H2Compensation

CO℃

0 〜 20.9 vol%0 〜 10000ppm

0 〜 650℃

OptionIR PrinterHT-1610Roll paper x 1AAbattery x 4

Printer paper
HT-1636

５rollsAmbient air
temperature sensor
HT-1321

O2 and CO, 2 measurement items
Compact design, 360ｇ

Long seller, loved by professional
maintenance person

CO with H2 Compensation
as standard functionMore accurate CO measurement by mean of H2 compensation
function.

Attach caseHT-1315

Made by aluminium
Size
350×460×155mm
2.9kg

Measurement
Software for PC
Online View 2000
HT-2084

RS232 transfer cable
USB cable
Recommended OS：
Windows7/8.1/10

Measurement
Software for PC
Online View 2000
HT-2094

RS232 transfer cable
Recommended OS：
Windows 7/8.1/10

Battery charge forabroad
HT-1318

AC100 〜 240V

Sampling probe
HT-1001A

L=300mm φ6mm
Inc.. condensate trap
Flue gas temp.;
0 〜 650℃

Probe handle
HT-7201A

Sampling hose length
1800mmInc. condensate trap

Probe tube
HT-7231

L=180ｍｍ φ5mm
Flue gas temp.;
0 〜 650℃

Probe tube
HT-7235

L=500ｍｍ φ6mm
Flue gas temp.;
0 〜 650℃

Probe tube
HT-7232

L=750ｍｍ φ6mm
Flue gas temp.;
0 〜 650℃

Probe tube
HT-7233

L=750ｍｍ φ8mm
Flue gas temp.;
0 〜 1100℃

Integrated data logger function
100 data storage is possible

0 〜 100℃
For air temperature

16

Speciﬁcation

Measurement items

Standard equipment

Measurement range
Accuracy
Resolution
Response time
Measurement range
Accuracy

O2＊1CO（H2Conpensation）

Measurement range
Accuracy0 〜 20.9 vol%±0.2 vol%0.1 vol%<10sec.
0 〜 10000ppm
0 〜 200ppm：±10ppm
200ppm 〜：±5％ reading
1ppm<40sec.
0 〜 650℃
0 〜 1100℃（depend on probe tube）±1℃ or±1% reading（0 〜 650℃）＊2±2% reading （650.1 〜 1100℃）
0.1℃（0 〜 999.9℃） 1℃（1000 〜 1100℃）
0 〜 100℃± 1℃

Resolution

0.1℃

Resolution
Response time
Measurement range
Accuracy

Flue gas temp.

Resolution
Ambient air temp.＊3

Calculated items＊4
CO2
CO calculated from O2 CO（O2＝0％）
CO calculated from O2 CO（O2＝？％）
Air ratio
Losses
Eﬃciency
Dew point

0 〜 CO2Max（calculated from O2）
0 〜 calculated ppm（Fixed O2 value at ０％）
0 〜 calculated ppm（User selectable O2 ％）1.0 〜 50.0
0 〜 100％
0 〜 100％0 〜 100℃

SensorTemperature
Display
Dimension
WeightBatteryOption
Probes
HT-1001A L=300mm φ6mm / with condensate trap 0 〜 650℃
Sampling probe
HT-1006L L=120mm φ5mm / with condensate trap 0 〜 650℃
Ｌ-Probe
Probe handle
HT-7201A with hose, condensate trap
Probe handle
Probe tube（Probe handle HT-7201A）
HT-7231 L=180ｍｍ φ5mm 0 〜 650℃
Probe tube
HT-7232 L=750ｍｍ φ6mm 0 〜 650℃
（With ﬂue gas temp. sensor）
HT-7233 L=750ｍｍ φ8mm 0 〜 1100℃
Temp. probe
Kthermocouple HT-1251a φ3×130L, 0〜950℃,for air/liquids
HT-1252a φ1.5×130L, 0〜950℃,
for air/liquids
HT-1253a φ3×130L, 0〜400℃,
for air/liquids/food, centric top
HT-1254a 130L, 0〜400℃,
for surface /rifts/air/liquid
HT-1255a φ4×130L, 0〜650℃,

Air temp.probe
Pt2000Ω
Ambient air temp. sensor Pt2000Ω
Battery charger
Hard case
Measurement software（with RS232cable）
HT-2084
Measurement software（with RS232 cable・USB conversion cable）
HT-1610
Infrared printerHT-1636
Roll paper for printerHT-1376
Hood

13A、6C、LPG、kerosine、Light oil、Banker A、Banker Ｃ、Pellet

Main unitPump, Interface for PC (RS232)
Data logger (possible to store 100 data), Interface for printerAccessory
Battery, Sampling probe, Condensate trap, Reﬂection plate, Carrying case, Softcase, Filter set
Art. No.
Spec. / remarks
Description
Sampling probe( with ﬂue gas temp. sensor) HT-1006 L=180mm φ5mm ,with condensate trap, ﬂue gas temp.0 〜 650℃

HT-1256a
HT-1257a
HT-1382
HT-1321
HT-1318
HT-1315
HT-2094

FuelO2CO（H2Compensation）
Flue gas temp.
Ambient air temp.

Integrated

Electrochemical sensor
Electrochemical sensor
K thermocouple
Pt2000Ωfor surface /air/liquid0〜450℃, magnetic probe for surface
0〜180℃, Pliers probe for plates, tube
0〜100℃, for air
0〜100℃
AC100〜240V
Aluminum, dimension : 350×460×155,weight : 2.9kg
Online View 2000
（Recommended OS：Windows 7/8./1/10）
Online View 2000
（Recommended OS：Windows 7/8./1/10）
Roll paper ×1 AA battery× 4
5 rolls
for hot water supply device (gas/oil), stove＊1：Compensate the measured value when measure ﬂue gas include H2.
Operation：0℃ 〜 ＋45℃
Storage：−20℃ 〜 ＋60℃
＊2：The higher value is valid.
Dot matrix, 4lines＊3：Ambient air temp. (optional items) sensor is needed.
（W×H×D）80×150×35 mm＊4：Depend on area ﬂue gas may have diﬀerence, so calculated data also may have diﬀerence.
Approx. 360g
＊Accuracy is based on cylinder gas.
Battery charger（AC100V 50/60Hz DC12V 100mA）
Rechargeable NiMH battery（Approx. 8 hours continuous working hours ）Speciﬁcation might be changed without notice.

HT-1300N
Standard set
・Main unit・Sampling probe HT-1006
・Battery charger
・Carrying case・Soft case・Reﬂection place
・Filter set・User manualPrice 198,000 yen（tax exclude）

HT-1300N
Probe tube
300mm set・Main unit・Sampling probe HT-1001A・Battery charger
・Carrying case・Soft case・Filter set
・User manualPrice 199,000yen（tax exclude）

17

Flue gas analyzer

HT-1300N
Dimension

φ61.6

Inlet port gas

Charge LED

35

Battery charger port
Reﬂection plate

Interface RS232
13.0

φ5.2

Ambient temp. sensor port

φ32

IR printer Interface

KφdL

Flue gas inlet port
Flue gas temp. sensor

Sampling hose
FilterConeCondensate trap

Art.No. L(mm) K(mm) φd(mm) Sampling hose length (mm) Flue gas temp.（℃）
50 〜 650
HT-1006 0 〜 180 290
1800

80
Flue gas temp. connector
Flue gas temp. compensation cable

OptionSampling probeKL
φdDescription
Sampling probe

150

169

Silica gel

Flue gas inlet port
Flue gas temp. sensor
Cone
Flue gas temp. compensation cable
Sampling hoseFilter

silica gel

Condensate trap
Description
Sampling probe

Art. No. L(mm) K(mm) φd(mm) Sampling hose length(mm) Flue gas temp.（℃）
HT-1001A 0 〜 300 410
1800
60 〜 65018