Instruction Manual

PROFESSIONAL

ModelNo.

DIAL CALIPER GAUGE

Outside：AO- * /BO- *Inside：AI- * /BI- * /CI- *● Easy to carry・Fast, one touch operation● Shock proof mechanism・Dial clamp● 1.6mm ball measuring tip (AI-0, BI-0 have 1mm hardened,sharp tip. CI-* have 0.6mm hardened knife edge tip)

● For safe and proper use, please read this instruction

SAFETY PRECAUTIONS■ USE● Outside - Measure groove
diameter and thickness ofodd shapes (AO・BO)● Measure deep holes, groove
ID, widths（ AI・BI・CI）

SAFETY NOTIFICATIONS

manual before use and follow the procedures
described. Please keep manual where it isaccessible to user for future reference.
● Keep this manual with the instrument if transferredor leased to a third party.
● For inquiries about this product, please contact
distributor or place of purchase.

Risk of PERSONAL INJURY orPROPERTY DAMAGE if not followed.
PROHIBITED action.
REQUIRED step or necessary condition.

Please Observe

To prevent harm to yourself or others, and to prevent damage to property, always follow theprocedures marked with the symbols shown below.
■ Procedures are marked to indicate the degree of harm which could occur if procedure is not followed:

WARNING Indicates risk of personal injury or property damage if not followed.
■ Procedures necessary for proper and safe care are shown with the following symbols:

Denotes a prohibition – You MUST NOT do

Denotes a requirement – You MUST do

WARNING
Read the manual and follow the directions.

・Use of product other than as described inthis manual may cause accident.

Use only for precision measuring.

・Use for any purpose other than measuringmay damage or wear the instrument.
Improper use may also cause accident.

Handle with Care.

・Do not drop or subject to excessive
shock. Do not place under heavy objects.
Improper handling may cause damage orpoor accuracy.

※ Model AO-1 shown.

Dial Clamp

Fixed Arm

Rev. Counter
Scale

Thank you for purchasing Niigata Seiki Dial Caliper Gauge.
This product is a precision gauge for measuring a dimension at the end of the two arms, either
inside (internal) or on the outside (external) on a workpiece.

■ FEATURES

Outside Measurement 【AO- * / BO- * 】

Use in environments meeting the following
conditions:● Do not expose to dust, oil, moisture, or
direct sunlight.
● Protect from use by children and unauthorizedpeople.
・Use in places contrary to the above may affectaccuracy or cause damage to the productresulting in accident or injury.

Do not disassemble or modify.

・ It may damage product or cause accident.

PointerMovableArm

LeverBezel

1. Hold the gauge in your hand and squeeze the Lever toopen the Arms.
2. Set the gauge by measuring a master work piece orgauge block, and rotating the Bezel to align the dial
face with the Pointer.
3. Squeeze Lever to open, and insert the part to bemeasured. Gently release the Lever and read the Scale
making sure the Arms are positioned to measure thedesired dimension.
※ Make sure the arms are not tilted with respect to theaxis of the dimension to be measured.

Inside Measurement 【AI - * / BI - * / CI - * 】※ Model AI-2 shown.

Push Button
PointerRev. Counter

Fixed Arm

Scale
MovableArmDial
ClampBezel

MODEL SPECIFICATIONS【Outside】

units：mm

RepeatMeasurement
Accuracy ability Weight
Model GraduRangeation
80 g
0 〜 12
AO -1
83 g
10 〜 22 ±0.04
0.02
AO -2 0.01
85 g
20 〜 32
AO - 380 g
0 〜 25BO -1
83 g
BO -2 0.025 10 〜 35 ±0.075 0.05
85 g
20 〜 45
BO - 3●Body Material ： Steel

AFTER USE CARE, STORAGE
Remove any dust or dirt after use.
※Do not lubricate.

・Wipe any contamination from Arms
and sliding surfaces using a dry cloth,or cloth moistened with alcohol.
・To clean other surfaces, wipe with asoft dry cloth, or a cloth moistened witha mild cleaner.

Store in provided case in a cool, dark,and dry location.

・ During storage, make sure there is noforce on the Arms (such as pushed in,or lateral force.)
・ Keep away from moisture and direct
sunlight, and secure from unauthorized
personnel.

1. Hold the gauge in your hand and press the Button toopen the Arms.
2. Set the gauge by measuring a master work piece or ringgauge and rotating the Bezel to align the dial face with
the Pointer.
3. With the Button pressed, insert the gauge into theposition to take a measurement. Gently release theButton and read the Scale.

【Inside】

units：mm

Gradu- Meas.
Range

Model ation

3〜 9

A I-0
A I-1
A I-2

0.01

B I -1

10〜2220〜3230〜4240〜52

A I-3
A I-4
A I-5
B I -06〜180.02

10〜3410〜35

B I -2
B I -3B I -4
B I -520〜450.025
30〜55
40〜6550〜75

C I -05〜15

C I -1AC I -1BC I -2

0.01

10〜20
10〜30
20〜40

Arm
Length1530
2040
20404060
804080
80103030506060

Range
3〜 55〜 96〜 9
9〜18
10〜16
16〜2220〜32
30〜4240〜52
10〜2020〜3410〜2020〜3520〜45
30〜55
40〜65
50〜75
5〜 88〜15
10〜20
10〜20
20〜30
20〜40

RepeatAccuracy ability Weight

75 g
±0.04

0.02

75 g
80 g
73 g
85 g
88 g

±0.06

0.04

80 g
83 g

±0.075

0.05

±0.03

0.02

±0.05

0.02

85 g
88 g
90 g
93 g
85 g
85 g

95 g
95 g

To maintain measurement accuracy, periodic calibration is recommended.
Wear of measuring surfaces from repeated use may affect accuracy andperiodic accuracy checks should performd.
Outside Japan: Please contact distributor or place of purchase to inquire
about calibrations services.

Niigata seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

業務用

取扱説明書

外側用：AO- / BO-

品 番 内側用：AI- */ BI- /*CI***

ダイヤルキャリパゲージ

この度は「ダイヤルキャリパゲージ」をお買上げ頂きありがとうございます。
この商品は２本のアームを使って、外径測定または穴の直径や特殊形状の内径測定をする高精度な測定器です。

■特長●小型軽量・ワンタッチ操作で作業時間の短縮に●ショックプルーフ機構・ダイヤルストッパ付●測定子1.6mmボール（AI - 0・B I - 0は先端 1 mmシャープ
エッジ焼入れ、C I -＊は先端0.6mmナイフエッジ焼入れ）●正しく安全にお使いいただくため、ご使用の前に本取扱説明書を必ず読み、記載の手順に従ってご使用ください。
お読みになった後は、お使いになる方がいつでも見ることができる場所に、大切に保管してください。
●第三者に譲渡・貸与される場合も、この説明書を必ず添付してください。
●当商品に関するお問い合わせは、お買い求めの販売店もしくは当社にご連絡ください。

安全上のご注意■用途●外 径・リ ン グ 溝 径・変 形
物外径測定に（AO・BO）●内径・深穴・溝内径・幅測定に（AI・BI・CI）

安全上のお知らせこの説明書には拡大損害が予想される事項には、禁止事項には、必ずお守り頂きたい強制事項にはマークを付けて表示しています。

お使いになる人や他の人への危害、財産への損害を未然に防止するため、必ずお守りいただくことを下記のように説明しています。
■表示内容を無視して、誤った使い方をした時に生じる危害や損害の程度を次の表示で区分しています。

注意 『傷害を負う、または物的損害が発生するおそれがある内容』です。

■お守りいただく内容の種類を次の図記号で区分しています。
必ず実行していただく内容『強制事項』です。

注意
取扱説明書をよく読み、指示に従う。

・取扱説明書に記載された内容以外での使用は、事故の原因となります。

精密比較測定作業にのみ使用する。

・指定用途以外へのご使用は、製品の破損や磨耗、予測できない事故の原因となります。

本器は大切に取り扱う。

・落下などの衝撃を与えたり、重量物を乗せたりしないでください。精度不具合や故障、破損の原因となります。

ダイヤルストッパ １．
ダイヤルキャリパゲージを軽く持ち、指でレバーを※イラストはAO-1です。

押すと、移動アームが広がります。
２．被測定物の基準寸法をマスタゲージ等で測り取り、べゼルを回転させて長針用目盛板を長針に合わせます。
長針用目盛板
３．レバーを押して被測定物に当て、静かに離し、測定子が正しく測定箇所に当たっていることを確認しなが長針ら目盛を読み取ります。
べゼル（リング） ※測定軸に対してアームが傾斜しないように注意して測定してください。

固定アーム

移動アーム短針レバー

内側測定方法 【AI - */ BI - */CI - * 】※イラストはAI-2です。

押ボタン長針短針

固定アーム

長針用目盛板

必ずお守りください。

してはいけない内容『禁止事項』です。

外側測定方法 【AO - */ B O - * 】

下記の条件を満たす環境で使用する。

●埃、油、オイルミストが少なく、直射日光が当たらない場所●子どもや、使用者以外が近付かない場所
・上記に反する場所での使用は、精度不良や製品の破損、事故やけがの原因となります。

分解、改造を行わない。

・精度不具合や製品の破損の原因となります。

移動アーム

ダイヤル
ストッパ

べゼル（リング）

本体仕様

【外側用】
品番

目 量

AO -1
AO -2
AO - 3BO -1
BO -2
BO - 30.010.025

測定範囲

単位：mm繰返し指示誤差 精密度 質 量0 〜 1210 〜 22 ±0.04
20 〜 320 〜 2510 〜 35 ±0.075
20 〜 45●本体材質：スチール

0.020.05

80 g
83 g
85 g
80 g
83 g
85 g

使用後のお手入れ・保管方法
使用中に付着した埃、汚れを清掃する。
※注油は行わないこと。
・アーム摺 動面の 汚 れは、乾いた布かアルコールを少量 含ませた布で拭き取ってください。
・その他外観の汚れは、柔らかく乾いた布か、中性洗 剤を少量含ませた布で拭き取ってください。

付属のケースに収納し、乾燥した冷暗所に保管する。

・保管時、アームに負荷をかけないよ
う、力（押し込み力・曲げ荷重など）をかけないでください。
・直射日光や湿気は避け、管理者以外が触れない状態で保管してください。

１．ダイヤルキャリパゲージを軽く持ち、押ボタンを押すと移動アームが固定アームの方向に移動します。
２．被測定物の基準寸法をマイクロメータ、リングゲー
ジ、マスタゲージ等で測り取り、べゼルを回転させて長針用目盛板を長針に合わせます。
３．押ボタンを押しながら被測定物に挿入し、静かに押ボタンをはなし、目盛を読み取ります。

【内側用】
品番

目 量 測定範囲

A I- 03〜 9

A I- 1
A I- 20.01

B I -1

10〜2220〜3230〜4240〜52

A I- 3A I- 4A I- 5
B I -06〜180.02

10〜3410〜35

B I -2
B I -3B I -4
B I -520〜450.025
30〜55
40〜6550〜75

CI -05〜15

C I -1AC I -1B
CI - 20.01

10〜2010〜30
20〜40

測定可能 測定可能
指示誤差
囲
アーム長さ 範153020402040406080408080
103030506060

3〜 5
5〜 9
6〜 99〜1810〜1616〜2220〜32
30〜4240〜52
10〜2020〜3410〜2020〜3520〜4530〜55
40〜65
50〜75
5〜 88〜1510〜20
10〜2020〜30
20〜40

単位：mm
繰返し
精密度 質 量

75 g
±0.04

0.02

75 g
80 g
73 g
85 g
88 g

±0.06

0.04

80 g
83 g

±0.075

0.05

±0.03

0.02

±0.05

0.02

85 g
88 g
90 g
93 g
85 g
85 g

95 g
95 g

測定精度を保つため、定期校正を実施することをおすすめします

性能の劣化は、使用の頻度・環境によって大きく異なります。
お客様の社内規格に準じた周期で、定期点検を行うことをおすすめ致します。
当社にて校正を受け付けております。
全国にある当社代理店、取扱店を経由してお申し込みください。
発売元〒955-0055 新潟県三条市塚野目5丁目3番14号(0256)33-5502(代) FAX(0256)33-5528

URL http://www.niigataseiki.co.jp

E354-T2

21051000