COM-3010PRO
The negative ion measuring instrument for ores

The weak radiation generated from a natural ore or ceramics reacts to the molecule in the air, and an anion is made. A natural oreand ceramics are measured by a sensor, and it converts into the number of anions with the microcomputer of internal organs, anddisplays. Please use it as management of a product besides the cloth produced commercially using a natural ore and ceramics,
bedding, clothing, wall material, and accessories, and a measuring instrument for sales promotion.

Five kinds of measurement modes
1. Standard Measurement Mode
2. Speed Measurement Mode
3. Automatic Measurement Mode
4. Moving Average Measurement Mode
5 Investigation Mode

：20-second measurement：10-second measurement：Exact measured value：Somewhat exact measured value：Investigation of the source of a signal

Printer printingMeasurement mode
■ Mode 1: Standard measurement mode

■ When a cable is connected to a printer and the power supply is

It measures for 20 seconds. Measured value is calculated
to negative ion and a value is displayed on a display forindication.

on, measurement data is automatically printed after thecompletion of measurement.
■ At the time of automatic measurement mode, the datameasured 8 times and the data of the calculated maximum, theminimum, and average value are printed.

■ Mode 2: Speed measurement mode
It measures for 10 seconds. Measured value is calculated tonegative ion and a value is displayed.

■ The example of printing according to measurement mode
001: Standard Measurement Mode002: Speed measurement mode
003: Automatic measurement mode
004: Moving Average Measurement
Mode■ Mode 3: Automatic measurement mode
Measurement for 20 seconds is performed 8 times.
Themaximum, the minimum, and average value are calculated
and average value is displayed on a display for indication.

001-ION: 8601 ion/cc002

FAST: 8466 ion/cc003■ Mode 4: Moving average measurement mode
The measurement for 20 seconds is repeated 16 times for every
second. The number of negative ion averages 16 times ofmeasured value, and displays it on a display for indication.

■ Mode 5: Investigation mode

AUTO MEASUREMENT
1: 9015

2: 8799

3: 8904

4: 8793

5: 8952

6: 8772

7: 8910

8: 9180

MAX: 9180

MIN: 8772

AVE: 8915 ion/cc

The number of signals is counted up. When the source of asignal is strong, a count becomes early. When the source of asignal is weak, a count changes late.

004

M-AVE: 8900 ion/cc

A printer is an option.

Specification■ Display for indication： 4 figure+1 / 2 digital liquid crystal display■ Signal LED： It is LED lighting with a signal.
■ Time base range： 0-100,000 pieces/cc■ Printer output： It prints to an option printer.
■ Automatic printing： It is automatic printing at the time of the end of measurement.
■ Power supply： Four AA alkaline dry cells■ Accessories： A leather case, four AA alkaline dry cells, an operation manual.

■ Beep： A piezo-electric buzzer■ ×10 LED： It is LED lighting when increasing 10 times.
■ Detector： GM sensor■ Low battery ： Consumption of a dry cell will display LOBAT.
■ Auto-power-off： A power supply is shut off automatically.
■ Case weight： 420g (with no battery)■ Case size：140mm × 77mm × 53mm

Development manufacturer

COM SYSTEM,INC.
Postcode: 196-0003 2-3-17, Matsubara-chou, Akishima-City, Tokyo JAPANTEL:81-42-543-9062
FAX:81-42-543-9570
http://www.com-system.co.jp
E-mail: com@com-system.co.jp