Catalogue: mp6 accessories
Version 1.9 (Dec 2023)

Contents
1

Revision History............................................................................................................................... 32

mp6 series accessories .................................................................................................................... 3

3

4

2.1

Bubble Trap mp-bt .................................................................................................................. 3

2.2

Filter mp-f................................................................................................................................ 4

2.3

Passive Check Valves ............................................................................................................... 4

2.3.1

mp-cv............................................................................................................................... 4

2.3.2

sm-cv ............................................................................................................................... 5

2.4

Pulsation Damper mp-Damper ............................................................................................... 5

2.5

Reservoir Bundle ..................................................................................................................... 5

2.6

Tubing connector mp-y ........................................................................................................... 5

2.7

Tygon Tubing mp-t .................................................................................................................. 6

2.8

Silicone Tubing mp-s ............................................................................................................... 6

2.9

Hose Clamp mp-hc .................................................................................................................. 6

2.10

Takasago 2/2-way normally-closed valve ............................................................................... 7

2.11

Memetis 2/2-way bistable valve ............................................................................................. 8

2.12

Festo 3/2-way solenoid valve (not media separated) ............................................................. 8

2.13

Bürkert 3/2-way solenoid valve (media separated) ................................................................ 8

2.14

Liquid Flow Sensors SLF3s-0600F & -1300F by Sensirion ........................................................ 9

2.15

Gas Flow Sensor by Sensirion .................................................................................................. 9

2.16

Preassembled Pressure Sensor by Honeywell......................................................................... 9

2.17

Microfluidic Chips by microfluidic ChipShop ......................................................................... 10

2.18

mp6-con cable for mp-Labtronix ........................................................................................... 10

2.19

mp6-mol connector............................................................................................................... 10

Important Notices ......................................................................................................................... 11
3.1

Warranty ............................................................................................................................... 11

3.2

Warning, Personal Injury ....................................................................................................... 12

3.3

Declaration of conformity ..................................................................................................... 13

Company information ................................................................................................................... 13

Page 2 of 13

Catalogue: mp6 accessories
Version 1.9 (Dec 2023)

1 Revision History
Changes from v1.2 to v1.3
• New chapter: 1 Revision History added• New sub chapter: 2.10 Memetis 2/2-way bistable valve• Wetted Materials in 2.3.2 updated• Wetted Material in 2.9 updated• Fluidic connectors added in 2.9• Compatible drivers in 2.9 updated
Changes from v1.3 to v1.4
• New sub chapter: 2.5 Reservoir Bundle
Changes from v1.4 to v1.5
• New sub chapter: mp-s
Changes from v1.5 to v1.6
• New sub chapter: Microfluidic Chip by Micronit
Changes from v1.6 to v1.7
• New sub chapter: Gasflow Sensor by Sensirion• Photo added for sub chapter 2.17
Changes from v1.7 to v1.8
• New sub chapter: Festo 3/2-way solenoid valve (not media separated)• New sub chapter: Bürkert 3/2-way solenoid valve (media separated)• Chapter Warning, Personal Injury: Paragraph 4 extended• Correction: several order codes• Extended: Internal volume for mp-damper
Changes from v1.8 to v1.9
• Grammar mistakes corrected

2 mp6 series accessories
2.1 Bubble Trap mp-bt
A bubble trap to get rid of bubbles in your flow stream.

Dimensions

Technical specification:
mp-bt

Order code: mp-bt

Dimensions

17 x 19 x 10 mmMaterial

PPSU, PETMedia

Water based solutionsports

Ø 1,9mm suitable for 1,3mm mp-t

Internal volume

100 µlWater Entry Pressure:

1,9 bar

Necessary neg. pressure

100 – 200 mbar

Necessary accessories

mp6 @ 250Vpp/300Hz

Page 3 of 13

Catalogue: mp6 accessories
Version 1.9 (Dec 2023)

2.2 Filter mp-f
Depending on the particulate matter, the particle load has an influence on the performance of the
micropump. The higher the load the larger the probability of clogging and blockage of the fluid
channels. This can lead to a decline until stagnation of the flow rate of the micropump. We recommend
only to pump liquids with a particle size smaller than 50 μm. We recommend the use of filters if you
are pumping media that has a particulate load.
Technical specifications:
mp-filter liquid filter
Dimensions
Wetted material
Fluidic connectors
Filter porosity

Order code: mp-f
21 mm x 5.5 mm (length x wrench size)
0.82677 in. x 0.2165 in.
PP, PPSU, DELO AD840
barbed tube clip, length : 5.6 mm
for tubing with internal diameter: 1.3 mm
20 – 60 µm

2.3 Passive Check Valves
When the micropump is switched off, a back flow of the pumping medium, depending on a differential
pressure between in- and outlet is possible. In order to impede a back-flow Bartels Mikrotechnik offersa passive check valve integrated in stainless steel.
The connection of the valves with the micropump can be done easily with suitable tubing. The valve
should be placed between the micropump and the outlet reservoir. The valves are individually marked
regarding the free flow direction. Please note that the volume flow of the micropump will be influenced
by the check valve.

2.3.1

mp-cv

Dimensions:Technical specifications:
mp-cv check valve
Dimensions
Materials in contact with the
pumped media

Order code: mp-cv
21 mm x 5.5 mm (length x wrench size)
0.82677 x 0.2165 in.
silicone, stainless steel

Cracking pressure

barbed tube clip, length : 5.6 mm
for tubing with internal diameter: 1.3 mm
typical < 35 mbar

Max. back pressure

500 mbar

Typical leak rate

<20 µl/h for DI-water (at 500 mbar)

Fluidic connectors

Page 4 of 13

Catalogue: mp6 accessories
Version 1.9 (Dec 2023)

2.3.2 sm-cv
sm-cv check valve

Order code: sm-cv

Barbed1/16”

Wetted Material

Polypropylene, Viton, Stainless SteelOpening pressure

0.09 PSI

2.4 Pulsation Damper mp-Damper
This damper is designed to reduce the pulsation of fluids from micropumps to allow sensors to measure
more reliable. For example, calorimetric flow sensors.
Technical specification:
mp-damper pulsation damper

Order code : mp-damper

Material

PPSU (black/transparent), silicone

Dimensions

17 x 19 x5 mm

Number of in-/ outlets

2

In-/ outlet inner diameter

1,4 mmIn-/ outlet outer diameter

1,6 mmInternal voulume

133 µlOperating temperature+5°C up to 45°C

Wetted material

PPSU, silicone

2.5 Reservoir Bundle
This bundle is equipped with components realizing direct flow (i.e. closed-loop) and indirect flow (i.e.
pressure driven flow) applications. It can also be utilized as a damping volume for gas application
minimizing pulsation in combination with gas flow meters.
Reservoir Bundle
Included components

Order code : Reservoir
Vial, closure w/hole & septum, short dosing tip,long dosing tip, Male Luer to Barb connector (2x)Dimensions vial

22.5 mm x 46 mm

Dimensions short dosing tip

ID : 1.6 mm, OD : 1.83 mm, Length : 38 mm

Dimensions long dosing tip

ID : 1 mm, OD : 1.4 mm, Length : 50 mm

Male Luer to Barb connector

OD Barb : 1/16 inch

Operating temperature< 100°C
glass, stainless steel, PTFE, Silicone, metal(magnetic), PP

Wetted material

2.6 Tubing connector mp-y
The y-connector is suitable for a parallel operation of two micropumps.
Technical specifications:mp-y tubing connector

Order code: mp-y

Material

polypropylene (PP)
1.3 – 2.6 mm
0.512 – 0.1024 in.

for tubing inner diameters ofPage 5 of 13

Catalogue: mp6 accessories
Version 1.9 (Dec 2023)

2.7 Tygon Tubing mp-t
The mp-t is a tube made of Tygon material that is compatible with the inlet and outlet of the
micropump. The tube is available in two versions.
Technical specifications:mp-t tubing

Order code: mp-t ID 1.3 mm

Inner diameter

1.3 mmOrder code: mp-t ID 1.02 mm
1.02 mm

Outer diameter

3 mm

2.74 .mm

Wall thickness

0.85 mm

Sterilizable

Yes (autoclave or ethylene oxide)

Colortransparent

Packaging unit

1m

Suitable micropumps mp6-liq, mp6-gas, mp6-gas+

2.8 Silicone Tubing mp-s
The mp-s is a tube made of silicone material that is compatible with the inlet and outlet of the
micropump.
Technical specifications:mp-s tubing

Order code: mp-s

Inner diameter

1 mm

Outer diameter

3 mmWall thickness

1 mmSterilizable

Yes (autoclave or ethylene oxide)

Color

Transparent/turbid

2.9 Hose Clamp mp-hc
The hose clamp mp-hc serves as a connector aiding the micropump functionality at high pressure
applications, by preventing leakage at the micropumps fluidic connectors.
mp-hc host clamps

Order code : mp-hc

Material

PPSU

Dimensions in close state4 x 5,1 x 2 mm (width in tube direction)

Inner diameter (closed)

3 mmPage 6 of 13

Catalogue: mp6 accessories
Version 1.9 (Dec 2023)
This frame helps to mount the micropump mp6 via screws.
Dimensions:Technical specification:
mp-mf

Order code: mp-mf

Dimensions

33 x 18 x 4 mmMaterial

PLA, blue

Suitable screw

M1,6

2.10 Takasago 2/2-way normally-closed valve
An almost silent and low power active valve from the company Takasago. We cooperate with the German
representative BMT.

Technical specification:
Active valve

Order code : Active valve

Valve type2/2-way valve – normally closed

Model number

Operating temperature range

SMV-2R-AN1F
4 x 18,4 x 19 mm
0,62x 1,57 x 0,06 in.
5 – 40°C

Material

FPM, PPS

Input current

250 mA

Switching time

600ms

Power consumption

0,3 W

Compatible driver

mp-valvedriverT

Dimensions

Fluidic Connectors
1.6 mm barb connectors
For more information please check the data sheet of the manufacturer Takasago, whichis available at https://www.pumps-valves.eu/

Page 7 of 13

Catalogue: mp6 accessories
Version 1.9 (Dec 2023)

2.11 Memetis 2/2-way bistable valve
An almost silent and low power active valve on a very small footprint from the company Memetis. It does notconsume any power within the switching states.
Active valve

Order code : Bistable (BV1101)

Valve type2/2-way valve – bistable

Model number

BV1101

Dimensions

20 x 5 x 10.1 mm³

Operating temperature range

10 – 50°C

Material

Silicone, PEEK

Input current

500 mA

Switching time

40 or 130 ms

Energy per switching

75 mJ

Compatible driver

mp-valvedriverM

Fluidic Connectors
1.6 mm barb connectors
For more information please check the data sheet of the manufacturer Memetis, whichis available at https: www.memetis.com

2.12 Festo 3/2-way solenoid valve (not media separated)
Fast switching 3/2-way solenoid valve from the company Festo that is not media separated.
Active valve

Order code: Festo solenoid valve

Valve type3/2-way valve – normally closed (not media separated)

Model number

VOVK

Dimensions

6 x 42.8 x 15.5 mm³

Operating temperature range

5 – 50°C

Material

PBT, SteelPower consumption

0.5 W

Switching time

<13 ms

Input voltage

12 V

Compatible driver

mp-valvedriverS

Fluidic Connectors
1.5 mm barb connectors
For more information please check the data sheet of the manufacturer Festo, which isavailable at www.festo.com

2.13 Bürkert 3/2-way solenoid valve (media separated)
Fast switching media separated 3/2-way solenoid valve from the company Bürkert.
Active valve

Order code : Bürkert solenoid valve

Valve type3/2-way valve – normally closed (media separated)

Model number

Typ 6724

Dimensions

9 x 44.8 x 40 mm³

Operating temperature range

5 – 50°C

Material

EPDM, PEEKPower consumption

1W

Switching timeCa. 3 ms

Input voltage

12 V

Compatible driver

mp-valvedriverS

Fluidic Connectors
2 mm barb connectors
For more information please check the data sheet of the manufacturer Festo, which isavailable at www.buerkert.dePage 8 of 13

Catalogue: mp6 accessories
Version 1.9 (Dec 2023)

2.14 Liquid Flow Sensors SLF3s-0600F & -1300F by Sensirion
Sensors from the SLF3x series by Sensirion are thermal flow sensors measuring the liquid flow rate of the pump.

Technical specification:
Liquid Flow Sensor

Order code : Sensor SLF3S-0600F or SLF3S-1300F

Dimensions

48 x 15.5 x 8.9 mm3
up to ±2000 µl/min (SLF3S-0600F)
Liquid Flow sensing
up to ±40 ml/min (SLF3S-1300F)
Wetted materials:
PPS, stainless steel 316L, epoxy-based adhesive
More information in the data sheet from the manufactuer Sensirion athttps://sensirion.com/

2.15 Gas Flow Sensor by Sensirion
Sensors from the SDP81x series by Sensirion are thermal flow sensors measuring the air flow rate of the pump.

Technical specification:
Gas Flow Sensor

Order code : Gasflow Sensor

Dimensions

29 x 10.25 x 18 mm3

Air Flow sensing

1 – 50 ml/min

Calibrated for

Air, N2PBT (polybutylene terephthalate), glass (silicon nitride,
Wetted materials:
silicon oxide), silicon, gold, FR4, silicone as static
sealing, epoxy, copper alloy, lead-free solder
Allowable overpressure
1 bar
More information in the data sheet from the manufactuer Sensirion athttps://sensirion.com/

2.16 Preassembled Pressure Sensor by Honeywell
The pressure sensors from the ABP series by Honeywell offer an easy-to-use solution measuring
pressure in liquid and gas systems.
Preassembled pressure
sensorDimensions
Pressure range

Order code : ABP pressure sensor
8 mm x 7 mm
Up to 15 psi

Measures gage pressure
More information in the data sheet from the manufacturer Honeywell available athttps://www.honeywell.com/us/enPage 9 of 13

Catalogue: mp6 accessories
Version 1.9 (Dec 2023)

2.17 Microfluidic Chips by microfluidic ChipShop
Plastic chips with internal fluidic structures helping to realize experiments on a very small footprint.

Technical specification:
Chip with 16 straightchannels

Chip with 7 reaction
chamber

Reaction chamberchip with 8 different
depthsWidth1000 µm--

Length

18 mm

-

-

Depth

200 µm370 µm100 – 800 µm

Volume3.6 µl6.66 µl

1.8 – 14.4 µl

Microfluidic
Chips:

Fluidic
interface

Mini Luer

Material

Topas

2.18 mp6-con cable for mp-Labtronix
mp6-con connection cable Order code: mp6-con- Molex FCC 1.25 mm pitch
Design and connectors- 85 cm (33.465 in.) cable- Binder 620 connector

2.19 mp6-mol connector
mp6-mol connector

Order code: mp6-mol

Connector to micropump mp6-series for custom made cabling

Contacts

Molex FCC 39532045
1.25 mm pitch
4

Entry Angle/Orientation

VerticalPC Tail Length

3.50 mm

Operating temperature-20°C – 80°C

Voltagemax. 200V

Current

max. 1.0 A per contact

Termination Interface: Style

Through HoleType

Page 10 of 13

Catalogue: mp6 accessories
Version 1.9 (Dec 2023)

3 Important Notices
This operating manual contains necessary instructions for the installation, commissioning, operationand maintenance of the mp6-series. The manual is intended to help you achieving optimal results in ashort time and shall also assist avoiding possible sources of errors. The operating manual of thecontrollers and the accessories are available separately.
The products have been designed with state-of-the-art technology and in accordance with all relevant
safety regulations. However, a risk of damage to the units, other property, the operator and/or otherpersons cannot be fully excluded.
Always ensure that specialized and trained personnel will comply with the following general
instructions. Therefore, please keep this manual and hand out copies as required.
All values are approximate and no guarantee of specific technical properties.
Changes in the course of technical progress are possible without notice.

3.1 Warranty
The mp6 micropumps have been developed for the transport of gases or liquids. The controllers havebeen developed for operating the mp6 micropumps. Bartels Mikrotechnik can assume no liability fordamages resulting from the pump media. This applies especially for hazardous fluids.
The mp6 micropumps must be operated with Bartels Mikrotechnik electronics. Bartels MikrotechnikGmbH cannot guarantee the proper work of the units with customer specific electronics. If othercontrollers than the ones from Bartels Mikrotechnik are used, Bartels Mikrotechnik disclaims any
warranty.
Bartels Mikrotechnik assumes no liability for abnormal handling, improper or negligent use of the mp6
micropumps and the controller that is not conform to the specified purpose of the system. This applies
especially for micropump controllers, components and systems of other manufacturers, which havenot been certified by Bartels Mikrotechnik.
We guarantee that the mp6 micropumps comply with the actual state of scientific and technicalknowledge and due to this, the operational risks are limited to a minimum.
Bartels Mikrotechnik GmbH warrants solely to the original purchaser of this product for a period of 12months (one year) from the date of delivery that this product shall be of the quality, material andworkmanship defined in Bartels Mikrotechnik GmbH published specifications of the product. Withinsuch period, if proven to be defective, Bartels Mikrotechnik GmbH shall repair and/or replace thisproduct, in Bartels Mikrotechnik GmbH’s discretion, free of charge to the Buyer, provided that:• notice in writing describing the defects shall be given to Bartels Mikrotechnik GmbH within
fourteen (14) days after their appearance• such defects shall be found, to Bartels Mikrotechnik GmbH’s reasonable satisfaction, to havearisen from Bartels Mikrotechnik GmbH’s faulty design, material or workmanship• the defective product shall be returned to Bartels Mikrotechnik’s factory at the Buyer’s
expense• the warranty period for any repaired or replaced product shall be limited to the unexpired
portion of the original period.
Page 11 of 13

Catalogue: mp6 accessories
Version 1.9 (Dec 2023)
This warranty does not apply to any equipment which has not been installed and used within thespecifications recommended by Bartels Mikrotechnik GmbH for the intended and proper use of theequipment.
EXCEPT FOR THE WARRANTIES EXPRESSLY SET FORTH HEREIN, Bartels Mikrotechnik GmbH MAKES NO
WARRANTIES, EITHER EXPRESS OR IMPLIED, WITH RESPECT TO THE PRODUCT. ANY AND ALL
WARRANTIES, INCLUDING WITHOUT LIMITATION, WARRANTIES OF MERCHANTABILITY OR FITNESSFOR A PARTICULAR PURPOSE, ARE EXPRESSLY EXCLUDED AND DECLINED. Bartels Mikrotechnik GmbHis only liable for defects of this product arising under the conditions of operation provided for in thedata sheet and proper use of the goods. Bartels Mikrotechnik GmbH explicitly disclaims all warranties,
express or implied, for any period during which the goods are operated or stored not in accordancewith the technical specifications. Bartels Mikrotechnik GmbH does not assume any liability arising outof any application or use of any product or circuit and specifically disclaims any and all liability,
including without limitation consequential or incidental damages. All operating parameters, including
without limitation recommended parameters, must be validated for each customer’s applications bycustomer’s technical experts. Recommended parameters can and do vary in different applications.
Bartels Mikrotechnik GmbH reserves the right, without further notice, (i) to change the productspecifications and/or the information in this document and (ii) to improve reliability, functions anddesign of this product.

3.2 Warning, Personal Injury
Bartels Mikrotechnik GmbH rejects any responsibility for damages to persons or property resulting
from non-compliance with the instructions in this manual. In this case all warranties shall be void.
Do not use this product as safety or emergency stop devices or in any other application where failureof the product could result in personal injury. Do not use this product for applications other than its
intended and authorized use. Before installing, handling, using or servicing this product, please consult
the data sheet and application notes. Failure to comply with these instructions could result in death orserious injury. If the Buyer shall purchase or use Bartels Mikrotechnik GmbH products for any
unintended or unauthorized application, Buyer shall defend, indemnify and hold harmless Bartels
Mikrotechnik GmbH and its officers, employees, subsidiaries, affiliates and distributors against all
claims, costs, damages and expenses, and reasonable attorney fees arising out of, directly or indirectly,
any claim of personal injury or death associated with such unintended or unauthorized use, even if
Bartels Mikrotechnik GmbH shall be allegedly negligent with respect to the design or the manufactureof the product.
Moreover, please note that components of the controller and pump are operating with high voltage.
Therefore, persons wearing pacemakers are recommended to avoid the operating system. Do notopen the housing of the micropump and the controllers.
The use of liquids, which may alone or in combination create explosive or otherwise healthendangering conditions (including vapors) is not permitted.
All work in connection with the installation, assembly, commissioning/decommissioning, disassembly,
operation, servicing, cleaning and repairing of the pump and the controller must be carried out by
qualified, suitably trained and instructed personnel. Work on electrical components and assemblies
must be carried out by supervised personnel with the necessary qualifications and skills, after verifyingthat the components do not lead residual voltage or charge and in a dry environment.

Page 12 of 13

Catalogue: mp6 accessories
Version 1.9 (Dec 2023)
We guarantee that the micropumps comply with the actual state of scientific and technical knowledge
and due to this, the operational risks are limited to a minimum.
ESD Precautions: The inherent design of the active electronic components, i.e. mp-Lowdriver, Highdriver, -Highdriver4, Valvedriver and -Labtronix causes it to be sensitive to electrostatic
discharge (ESD). To prevent ESD-induced damage and/or degradation, take customary and statutory
ESD precautions when handling this product.

3.3 Declaration of conformity
Bartels Mikrotechnik GmbH declares that the products are compliant to the RoHS directive
2011/65/EU. The controllers comply with the requirements of EMV 2014/30/EU and CE markings havebeen affixed to the devices. Additionally, the controllers are also compliant to the EU Low Voltage
Directive 2014/35/EU.

4 Company information
Bartels Mikrotechnik is a globally active manufacturer and development service provider
in the field of microfluidics. In the microEngineering division, the company supports
industrial customers in the modification, adaptation and new development of highperformance and market-oriented product solutions through the innovative means of
microsystems technology. The second division, microComponents, produces anddistributes microfluidic products and systems, especially for miniaturized and portableapplications. Our key products are micropumps that convey smallest quantities of gases or
liquids and are used in a variety of ways in biotechnology, pharmaceuticals, medical
technology and numerous other applications.
Bartels Mikrotechnik with passion for microfluidics!

Contact us:
Bartels Mikrotechnik GmbH
Konrad-Adenauer-Allee 1144263 Dortmund Germany

www.bartels-mikrotechnik.de
info@bartels-mikrotechnik.de
Tel: +49-231-47730-500

Follow Bartels:

⽇本代理店: 株式会社 ASICON
Your local contact in Japan: ASICON Tokyo Ltd.
asicon-tokyo@asicon-tokyo.com
WWW.ASICON-TOKYO.COM

Page 13 of 13