XL Flex andXL Flex+ VideoProbes
Shed light on your inspections.

The XL Flex and XL Flex+ VideoProbes provide unmatched image quality
and powerful light output to boost probability of detection (POD) and speed
up workflow. Rugged durability lowers your maintenance costs. Stereo
measurement lets you evaluate defects in depth. And expert support
from Waygate Technologies gives you peace of mind. All in a lightweight,portable, and easy-to-use device. Now featuring 16GB of memory and anupgraded 5” LCD screen on the XL Flex+.

Get accurate results, fast.
Complete efficient, precise inspections with unmatched
image quality, plus a 5” LCD screen on the XL Flex+.
Believe it when you see it.
Reduce eyestrain with powerful light output and a 3X brighter
display in the XL Flex+.
Cut down on costs.
Lower your maintenance spending with durability you cantrust, even in the harshest inspection environments.
Evaluate defects in depth.
Stereo measurement provides accurate analysis as you work.
Stay flexible.
At just 4.3 lbs., the device is lightweight and portable, with awheeled carry-on case for easy transport.

Learn quickly.
The intuitive interface includes Menu Directed Inspection(MDI) 2.0, an application to guide users through inspections
and organize results for simpler reporting.
Experience true peace of mind.
Get readily available assistance from Waygate Technologies
experts, wherever you are.

Now with a 5” LCD screen forclear-cut inspections.
The new XL Flex+ display is 85% larger and 3X brighter, withcrystal clear image quality unmatched in the industry. Pairedwith advanced optics and powerful light output, that means
sharper images right on your device. Because when you cansee clearly, you improve POD, speed up your workflow, anddeliver results with confidence.

Hard copy user manual(optional)

4 hour battery(optional)

Magic clamp and swivel ball
(optional)

Handset holder - below handset(optional)

Rigidizer(optional)

Optical tip adapter case

Tube gripper(optional)

Included battery charger/ac adapter

Integrated insertion tube storage reel2

Technical specifications
Operating environment
Tip operating temp:
System operating temp:-25°C to 100°C (-13°F to 212°F) Reduced-20°C to 46°C (-4°F to 115°F) LCD requires
warm-up period below 0°C (32°F)-25°C to 60°C (-13°F to 140°F)

Relative humidity:

95% max, non-condensing

Environmental:

MIL-STD-810G

Tested to United States Department ofDefense Environment Tests Sections 506.5,

articulation below 0°C (32°F)

Storage temperature:

Waterproof:

Standards compliance and classifications

Insertion tube and tip to 14.7 psi (1 bar, 10.2

507.5, 509.5, 510.5, 511.5, 514.6, 516.6, 521.3
MIL-STD-461F

XL Flex: Tested to United States Departmentof Defense Electromagnetic Interference
RS103 - ABOVE DECKXL Flex+: Available upon request
Standards complianceGroup 1, Class A: EN61326-1

UL, IEC, EN CSA-C22.2:61010-1

m of H2 O, 33.5 ft of H2 O)
XL Flex: Tested to IP55
XL Flex+: Tested to IP65

SystemUN/DOT T1-T8
IP rating

XL Flex: Tested to IP55

XL Flex+: Tested to IP65

CameraWheeled case:54.9 x 34.6 x 23.6cm (21.61 x 13.62 x 29)

4.0 mm (0.157 in) diameter probes

System weight:

In case:Image sensor:

1/10 inch Color SUPER HAD CCD camera

Pixel count:

290,000 pixels

Housing:

Titanium

XL Flex: 6.50 kg (14.30 lb)

XL Flex+: 8.39 kg (18.5 lb)
Without case:

XL Flex: 1.73 kg (3.8 lb)

XL Flex+: 1.98 kg (4.37 lb)
Power (Lithium Ion battery)

8.4 V, 38 Wh (2 hour) 8.4 V, 75 Wh (4 hour)Power supply:

AC Input: 100-240 VAC, 50-60 Hz, 1.5 A Max

Construction:

Polycarbonate housings with integrated

Dimensions:
LCD monitor:

DC Output: 9.5 V, 6.0 A

Versalon™(JP) bumpers

9.53 x 13.34 x 34.29 cm (3.75 x 5.25 x 13.50 in)
XL Flex: Integrated transflective 9.40 cm

6.1 mm (0.242 in) and 8.4 mm (0.331 in) diameter probes
Image sensor:

1/6 inch Color SUPER HAD™ CCD camera

Pixel count:

440,000 pixels

Housing:

TitaniumSoftware
Operating system:

Real-time multi-tasking operating system

User interface:

Simple drop-down menu-driven operation

File manager:

Embedded file manager software(3.70 in) active matrix VGA color LCD withmatrix

VGA color LCD, Transmissive display.
Joystick control:

Button set:
Audio:
Internal memory:ThumbDrive Copy between USB and C:\Measurements:

StereoProbe and Comparison

MDI software (optional):

Provides user defined guided inspection

Creates DICONDE compatible inspection
files Creates MS Word® compatible

360° All-Way tip articulation withXpertSteer, bump gesture, menu accesssupporting: File & Folder creation, naming,deleting Store to internal Flash (C:\) or USB

XpertBright, 640 x 480 resolution

XL Flex+: Integrated 12.7 cm (5.0) active

Menu navigation using articulation joystickinspection reports

and navigation

Audio data:

PC compatible (.AAC) file format

Access user functions, measurement andImage control:

Invert, Zoom (5X digital) Image Capture andDigital zoom:

Continuous (5.0X)

Image formats:

Bitmap (.BMP), JPEG (.JPG)

Video format:

MPEG 4

Text annotation:

Built-in full screen text overlay generator

digital functions

Integrated 2.5 mm headset/microphonejack

XL Flex: 4 GB Flash memory
XL Flex+: 16 GB Flash memory

Data I/O ports:

Two USB® 2.0 ports

Video port:

VGA Video Out

Brightness control:

Auto and Variable

Illumination type:White LEDLong exposure:Via auto and manual mode

White balance:

Factory default or user defined

RecallSupports international language character
sets 100 text line capacity

Graphic annotation:

User placement of arrows

Articulation control:“Steer & Stay” articulation lock/finearticulation Tip “Home” return to neutral
forward-tip orientation Userselectablefine or coarse control XpertSteer

probe articulation offers quick steering

responsiveness for tight probe control –bump steering enables slight adjustments
to probe position
Software updates:

Field updateable via USB ThumbDrive

Languages:

English, Spanish, French, German, Italian,Russian, Japanese, Korean, Portuguese,Chinese, Polish3

Insertion tube length and diameter optionsCamera diameter

Insertion tube working length

4.0 mm (0.157 in)

2.0 m (6.6 ft)

3.0 m (9.8 ft)

6.1 mm (0.242 in)

2.0 m (6.6 ft)

3.0 m (9.8 ft)

3.5 m (11.48 ft)

4.5 m (14.7 ft)

6.0 m (19.7 ft)

8.4 mm (0.331 in)

2.0 m (6.6 ft)

3.0 m (9.8 ft)

4.5 m (14.8 ft)

6.0 m (19.7 ft)

8.0 m (26.2 ft)

8.0 m (26.2 ft)
10.0 m (32.8 ft)

Tip articulation
Insertion tube length

Straight tube

2.0, 3.0 or 4.5 m

Up/down – 160° min, Left/right – 160° min

6.0 m

Up/down – 140° min, Left/right – 140° min

10.0 mUp/down – 120° min, Left/right – 120° min

Note: Typical articulation exceeds minimum specifications

Tip optics
Tip view(DOV)

Tip color

Field of view

Depth of field(FOV)(1)4.0 mm optical tip(DOF)part #

6.1 mm optical
tip part #8.4 mm optical tip
part #

Standard tips
FORWARD

NONE

80˚

35 mm–infinity(1.38 in–infinity)

T4080FFFORWARD

BLACK115˚

4 mm–infinity(0.16 in–infinity)

T40115FNFORWARDNONE

50˚

50 mm–infinity(1.97 in–infinity)

FORWARD

WHITE

50˚

12–200 mm(0.47–7.87 in)

XLG3T6150FG

FORWARD

ORANGE80˚

3–20 mm(0.12–0.79 in)

XLG3T6180FNFORWARDYELLOW90˚

20 mm–infinity(0.79 in–infinity)

XLG3T6190FF

FORWARD

BLACK

120˚

5–120 mm(0.20–4.72 in)

XLG3T61120FG

PURPLE50˚

12–80 mm

(0.47–3.15 in)

XLG3T6150FBFORWARD

BLACK

120˚

5–200 mm

0.20–7.87 in)

XLG3T84120FNFORWARDNONE

40˚

250 mm–infinity(9.84 in–infinity)

XLG3T8440FF(2)FORWARDOBLIQUEXLG3T6150FF(2)FORWARD

WHITE

40˚

80–500 mm(3.15–19.68 in)

XLG3T8440FG

FORWARDYELLOW80˚

25–500 mm(0.98–19.68 in)

XLG3T8480FG

FORWARD

GRAY

120˚

20-infinity(0.79”-infinity)

SIDE

BLUE

120˚

6 mm–infinity(0.24 in–infinity)

T40120SF

SIDE

RED

115˚

1–30 mm(0.04–0.18 in)

T40115SN

SIDE

BROWN

50˚

45 mm–infinity(1.77 in–infinity)

SIDE

GREEN

50˚

9–160 mm(0.35–6.30 in)

XLG3T6150SG

SIDE

BLUE

120˚

4–100 mm(0.16–3.94 in)

XLG3T61120SG

SIDERED80˚

1–20 mm(0.04–0.79 in)

XLG3T6180SN

SIDE

BROWN

40˚

250 mm–infinity(9.84 in–infinity)

XLG3T8440SF(2)T61120FF

XLG3T6150SF

SIDE

GREEN

80˚

25–500 mm(0.98–19.68 in)

XLG3T8480SG

SIDE

BLUE

120˚

4–200 mm(0.16–7.87 in)

XLG3T84120SN

Stereoprobe measurement tips
FORWARDBLACK

55˚/55˚

5 mm–infinity(0.2 in-infinity)

FORWARDBLACK

60˚/60˚

4–80 mm(0.16–3.15 in)

FORWARDBLACK

60˚/60˚

4–50 mm(0.16–1.97 in)

SIDE

BLUE

55˚/55˚

4 mm-infinity(0.16 in-infinity)

SIDE

BLUE

50˚/50˚

2–50 mm(0.08–1.97 in.)

SIDEBLUE

60˚/60˚

4–50 mm(0.16–1.97 in)

TM405555FG
XLG3TM616060FG
XLG3TM846060FG
TM405555SGXLG3TM615050SG
XLG3TM846060SG

FOV is specified diagonally.
(2)
Indicates tips with maximum brightness.
(1)

Copyright 2020 Baker Hughes Company. All rights reserved.
BHCS34210 (04/2020)

waygate-tech.com