https://global.kikusui.co.jp/

AC/DC Power Supply

The never-ending evolution of power supplies!

Ultra-Compact AC/DC Programmable Power Supply

PCR-WEA/WEA2 SeriesCompact size: 6 kVA in 6U size (PCR6000WEA2)
Up to 36 kVA in one single unit
100% regenerative capability (for "R" models, PCR-WEA2R)
Mix-and-match parallel operation up to 144 kVA
Flexible digital interface: LAN (LXI), USB, RS232C, GPIB (option)
Power line disturbance simulation
Power-saving function
DC output (100% of rated power)
Output frequency up to 5 kHz
Output rating: AC 0 to 320 Vrms, DC 0 to ±452 VNEW

3 phase 400 V
input modelTHE EVOLUTION
More power, more speed, more freedom! While maintaining the high-power density of
6 kVA/6U and 36 kVA in a single housing unit, the maximum output voltage, response
characteristics, and load stability have been improved!

Ultra-Compact AC/DC Programmable Power Supply

PCR-WEA/WEA2 Series
The PCR-WEA/WEA2 is a series of multifunctional switching AC
power supplies that combines accurate, high-power output with anultra-compact design. The 15 model line-up ranges from 1 kVA
to 36 kVA AC/DC with single & 3-phase variable output
from 6 kVA and up. The PCR-WEA2 also features amm
262(10.32 inch)

regenerative mode*1 that can drastically reduce power
consumption and cut operating costs.
The PCR-WEA2/WEA2R also supports mix-and-match
parallel operation*2 up to 144 kVA for large-scale test
systems. Output frequency up to 5 kHz is also available with allmodels, which is critical for AC applications in avionic industries.

● Compact size: 6 kVA in 6U frame (PCR6000WEA2)● Up to 36 kVA in a single unit (PCR36000WEA2)*1● 100% regenerative-power capability● Mix-and-match parallel operation up to 144 kVA● Flexible digital interface:
LAN (LXI), USB, RS232C, GPIB (option)● Power line disturbance simulation features● Sequence function for advanced simulation● External analog, digital control function (standard)● Power-saving function● DC output (100% of rated power)● Output frequency up to 5 kHz● Output rating: AC 0 to 320 Vrms, DC 0 to ±452 V*3● Analog monitor output (factory option)*1: Only "R" models (PCR-WEA2R) with 3-phase 200 V/400 V input.
For regeneration within the installation site only.
*2: Parallel operation is available for 6 kVA models and up, with amaximum of 4 units. Same model combination is not required.
*3: 6 kVA models and higher

Single-phase

Multi-type

PCR1000WEA
PCR2000WEA

PCR3000WEA2

1 kVA / 2 kVA

2Max
voltageincreased
to

320 V

PCR6000WEA2PCR6000WEA2R
Refer to pg.16 for full scale.

Multi-type

18 kVA

Multi-type

12 kVA

Multi-type

6 kVA

3 kVA

PCR6000WEA2PCR6000WEA2R

PCR12000WEA2PCR12000WEA2R

PCR18000WEA2PCR18000WEA2R

● Lineup
Specifications

Model

AC mode output rating
Power
Phase voltage
capacity

Phase

VA

V

DC mode output ratingMax.
Frequen- Powercurrent *1
cycapacity(L/H range)
A

Hz

VoltageW

V

Input rating (AC rms)

Max.
current *2(L/H range)

Phase

Voltage (nominal)

Apparent
Current
power

V

kVA or less A or lessA

PCR1000WEA

Single-phase

1k

10/5

1k

10/5

Single-phase

100 to 120, 200 to 240

1.4

17/8.5

PCR2000WEA

Single-phase

2k

20/10

2k

20/10

Single-phase

100 to 120, 200 to 240

2.7

32/16

3k

30/15

Single-phase

100 to 120, 200 to 240

4

48/24

3-phase 3-wire 200 V

Line voltage 200 to 240

6k

60/30

3-phase 3-wire 400 V

Line voltage 380 to 480

Single-phase
PCR3000WEA2

30/15

3k

Three-phase
Single-phase Three-wire

10/5

2k

Single-phase
PCR6000WEA2R
PCR6000WEA2

60/30
6k

Three-phase

20/10

Single-phase Three-wire

4k

Single-phase
PCR12000WEA2R
PCR12000WEA2

120/6012 k

Three-phase
Single-phase Three-wire

8k

Single-phase
PCR18000WEA2R
PCR18000WEA2

18 k

Three-phase
Single-phase Three-wire

12 kSingle-phase
PCR24000WEA2R
PCR24000WEA2(The spec
guaranteedvoltage range)
1 to 160/
2 to 320(L/H output
range)(Voltagesetting range)
0 to 161.0/
0 to 322.0

PCR30000WEA2Single-phase Three-wire

PCR36000WEA2

1
to
5000

18 k

240/120
24 k(The spec
guaranteedvoltage range)±1.4 to ±226/±2.8 to ±452
(L/H output
range)(Voltagesetting range)-227.5 to+227.5/-455.0 to+455.0120/60180/90240/12016 k
300/150
30 k

Three-phase

100/50Single-phase Three-wire

30 k

300/150

20 k

Single-phase
PCR36000WEA2R

60/30

80/40

Single-phase
PCR30000WEA2R

180/9024 k

Three-phase

12 k

40/20

360/18036 k

Three-phase120/60Single-phase Three-wire

36 k

24 k

360/1803-phase 4-wire 400 V

Line voltage 380 to 480

3-phase 3-wire 200 V

Line voltage 200 to 240

3-phase 3-wire 400 V

Line voltage 380 to 480

3-phase 4-wire 400 V

Line voltage 380 to 480

3-phase 3-wire 200 V

Line voltage 200 to 240

3-phase 3-wire 400 V

Line voltage 380 to 480

3-phase 4-wire 400 V

Line voltage 380 to 480

3-phase 3-wire 200 V

Line voltage 200 to 240

3-phase 3-wire 400 V

Line voltage 380 to 480

3-phase 4-wire 400 V

Line voltage 380 to 480

3-phase 3-wire 200 V

Line voltage 200 to 240

3-phase 3-wire 400 V

Line voltage 380 to 480

3-phase 4-wire 400 V

Line voltage 380 to 480

3-phase 3-wire 200 V

Line voltage 200 to 240

3-phase 3-wire 400 V

Line voltage 380 to 480

3-phase 4-wire 400 V

Line voltage 380 to 480

7.8

27
14

15.6

53
28

23.4

80
42

31.2

106
56

39

133
70

46.8

159
84

*1 When the output phase voltage is between 100 Vac and 161 Vac or 200 Vac and 322 Vac, the output current is reduced by the output voltage. When the output frequency is between 1 Hz and 40 Hz,
the output current is reduced by the output frequency.
*2 When the output voltage is between 100 Vac and 226 Vac or 200 Vac and 452 Vac, the output current is reduced by the output voltage.

★ 500 Hz Limit Model is available. The PCR-WEA2 Series offers a limited frequency type with a maximum output frequency of 500 Hz.

Multi-type

30 kVA

Multi-type

36 kVA

FeaturesMulti-type

24 kVA

PCR24000WEA2
PCR24000WEA2R

PCR30000WEA2PCR30000WEA2R

PCR36000WEA2PCR36000WEA2R

P4-P5

Performance

P6

Applications

P7

Exterior Design

P10-P11Specifications

P12-P15

Option/Cable

P8-P9, P18-P19

3

PWM Inverter Type - Programmable AC Power Supply
The PCR-WEA/WEA2 Series brings new innovations to the power-electronics industry.
Compact Size!

100% Regeneration Capability, No Time Limit

Compared to our previous PWM models, the size of the PCR-WEA
has been drastically reduced by 60%. Efficiency has also been increased by approximately 7%, for an overall high efficiency of approximately 85%.

The PCR-WEA2R models are capable of 100% power regeneration.
The power regeneration feature is available with absolutely no reverse load flow time limit. (30% for PCR-LE/LE2)

100% Maximum
Regeneration No reverse load flow time limit

Approx.

64%
lighterApprox.

63%

PCR6000W2

smallersize*Regeneration is limited within installation site. Only available in "R" models(PCR-WEA2R) with 3-phase 200 V/400 V input.

Approx. 120 kg
Previous model
PCR6000W2

Required output capacity

Weight

6 kVA

PCR6000WEA2RApprox. 43 kg

PCR6000WEA2R

Power running

Indoorregeneration

Compared with 6kVA

Inverse load flow

6 kVA

Approx.

7%higherefficiency

Load

Approx. 78%

Previous model PCR6000W2

Approx. 85%

Efficiency

PCR6000WEA2R

PCR6000WEA2R

Extremely Power Dense 36 kVA Chassis
The PCR-WEA/WEA2 form factor has been significantly improved,
occupying the absolute minimum amount of precious space in yourtesting facility. The form factor is even further optimized in high power models.

550 mm● Installation area comparison (36 kVA)
The PCR-WEA/WEA2 is only 1/6th the size of the PCR-LE!

PCR
36000WEA2

LAN, USB, RS232C Standard Digital Interface
The PCR-WEA/WEA2 series includes a flexible digital interface forusers utilizing LAN, USB, and RS232C communication interfaces(GPIB factory option available). LAN connection is LXI compliant,allowing you to monitor and control your device wherever you arevia computer, smartphone, or tablet web browser. This feature isparticularly important when conducting critical AC tests in anechoic
chambers/shield rooms. Additionally, the PCR-WEA can be controlled directly with easy remote-control software for customers withlimitations in external communication.

● Wired LAN connection (optical cable)

Less than 1/6th of the installation
area of previous model required!
Remote control (tablet PC)

2580 mm

430 mm*When compared with previous model PCR6000LE

Optical media converter

PCR6000LE × 6 units● Weight comparison (36 kVA)
The PCR-WEA/WEA2 is approximately 80% lighter than the PCR-LE!

80%
weightreduction

WeightOpticalfiber cable
30 m
Ethernet adapter

PCR36000WEA2LAN* Figure may differ from
actual display

Approx. 840 kg

Previous model
PCR6000LE × 6 units● Wireless LAN connection

Approx. 170 kg
PCR36000WEA2When compared interms of 36 kVA

LANWireless LAN router

Low Ripple Noise
Achieves an extremely low switching noise for a PWM inverter-type
AC power supply, with ripple noise as low as 0.25 Vrms.
The PCR-WEA series even boasts similar noise performance with
the PCR-LE/LE2 linear amplifier power supply series. The compact,
high-power design of the PCR-WEA/WEA2 has been achieved withabsolutely no compromises to ripple noise performance.
4

LAN

PCR36000WEA2Remote control (tablet PC)* Figure may differ from
actual display

Output Frequency up to 5 kHz

DC Output 100% of Rated Power

It has a maximum output frequency up to 5 kHz for critical applicationsin the defense and avionics industries. The frequency performanceof the PCR-WEA allows for simulation of sharp voltage fluctuations
required for airborne electronic equipment testing. Furthermore, thecompact 6kVA/6U form factor allows for the easy preparation of an
automated, one-rack testing system without requiring a costly, specialized power source installation space.

The PCR-WEA/WEA2 series enables DC output up to 100% of the
AC rated power output.

DC output: 100% of AC output rating
Required output capacity

6 kVA

PCR6000WEA2

AC : 6 kVA

Load

DC : 6 kW

Up to 144 kVA with Parallel Operation
Parallel operation is available on all models by simply connecting an
optional parallel operation cable. This feature is available even among
different models for a wide range of high power.
*Same input voltage and wiring system required for 6 kVA models and higher.

Paralleloperation

144 kVA

Maximum

Approx.1040 mm(PCR36000WEA2 × 4 units)

Power Saving Mode *6 kVA models and higher●Sleep mode

If the PCR-WEA/WEA2 does notdetect output for a certain amountof time, the power unit will go into"sleep mode" and cut power consumption.
Sleep modeiszzz....... screen
displayed.

●Power-saving mode

The power-saving feature allows
the PCR-WEA to cut the costs ofoperation by drawing power fromonly the necessary power modules
required to reach the output setting.

[Example]
Only 6 kVA drawn from the 36 kVA modelOFF

Approx.1800 mm

OFF● Mix-and-match parallel operation (example)

OFF

72 kVA
6 kVA

12 kVA

18 kVASLEEP

OFF

36 kVAOFF● Connection diagram
Slave unit

Slave unit

Slave unit

Master unit

PARALLELOUT INPARALLELOUT INPARALLELOUT INPARALLELOUT INON

Parallel operation connection cable

Modular design allows for simple maintenance

Analog Monitor Output (Factory option)
Instantaneous voltage / current values ​​can be output as voltage waveforms without using a differential probe or current sensor.
It is also possible to output the instantaneous value of the power value.
*6 kVA models and higher
GAIN 1/100.0 [V/A]
Monitor output

Amp

AC voltage
0 Vrms to 3.6 Vrms

Shunt

Power Line Error Simulation
The PCR-WEA/WEA2 series can simulate various power line abnormalities such as power outages, voltage drops (dips) and voltage increases (pops). This feature is useful for the testing of power-source
switches and various electronic devices.

OutputLoad
Output AC current
0 Arms to 360 Arms

Example: PCR36000WEA2 single-phase
Output current waveform (waveform output)

Each separate power module can be removed and replaced formaintenance and calibration. *For models 6 kVA and higher* Analog monitor output can monitor up to 3
outputs by selecting either voltage or current.

Power outages

increased voltage (pops)

decreased voltage (dips)5Built-in parallel operation driver software! Easy parallel operation with a single connection cable.
The PCR-WEA/WEA2 series can be easily configured in a parallel connection with a single cable* per connection for all models 6 kVA andabove. This cable can be used in synchronization with a power-interlock cable* to control the ON/OFF status of master/slave units. *Optional

Performance●Example of the combined system using same models
Capacity

Model

Qty

Parallel operation cable

Qty

Power-sync cable

Qty

12 kVA

PCR6000WEA2

2

PC01-PCR-WE

1

LC01-PCR-LE

1

48 kVA

PCR24000WEA2R

2

PC01-PCR-WE

1

LC01-PCR-LE

1

90 kVA

PCR30000WEA2R

3

PC01-PCR-WE

2

LC01-PCR-LE

2

144 kVA

PCR36000WEA2R

4

PC01-PCR-WE

3

LC01-PCR-LE

3[PCR36000WEA2R 4 units, example of 144 kVA]
The figure below is a conceptual diagram. Power wiring etc. are also required for system build up. Please consult your local Kikusui distributor.
OUT

IN

Slave

OUT

IN

Slave

J1

J2

OUT

IN

Slave

J1

J2

Parallel-operation cable

MasterJ1

J2

Power-sync cable●Example of the combined system using different models
Capacity

Model

Part

Qty

PCR6000WEA2R

AC/DC Power supplies (6 kVA)1

PCR12000WEA2R

AC/DC Power supplies (12 kVA)

1

60 kVA

PCR18000WEA2R

AC/DC Power supplies (18 kVA)

1

Parallel-operation system

PCR24000WEA2R

AC/DC Power supplies (24 kVA)

1

PC01-PCR-WE

Parallel operation cable3

LC01-PCR-LE

Power-sync cable3

The figure below is a conceptual diagram. Power wiring etc. are also required for system build up. Please consult your local Kikusui distributor.
OUT

IN

OUT

IN

OUT

IN

Parallel-operation cable

24 kVA
18 kVA

Master
12 kVA
6 kVA

Slave

Slave

Slave
J1

J2

J1

J2

J1

J2

Power-sync cable6

Applications
For Standard Compliance Testing● Single-phase system● Three-phase system
Harmonic/
Flickeranalyzer

Three
phase

Singlephase

Line-impedance
network
LIN series *1

Line-impedance
network
LIN series *1

HARMONIC / FLICKER ANALYZER

KHA3000
ASSISTMEMORY

MODE
HA

Vf

UTILITY

GPIB

OTHER

HARD COPY

FILE

HOLD

EXT CONTSYSTEM

REMOTELOCAL

KEY LOCK
VIEW

START

F1
F2
USB

F3
F4

REMOTE

F5
F6

STOP

ENTRY
9

SHIFT

7

8

V RANGE
4

I RANGE
5

6

V RANGE
1I RANGE
23ESC

BUZZER
0

.

-

ENTER

BUZZERBACKLIGHT

POWER

IEC dip
simulator
DSI series *1

Harmonic/Flicker
analyzer

LINE IMPEDANCE NETWORKLINE IMPEDANCE NETWORK

LIN3060J
ALARM

ALARM

HARMONIC / FLICKER ANALYZER

KHA3000
ASSISTMEMORY

MODE
HA

Vf

UTILITY

IMPEDANCE SELECTIMPEDANCE SELECT

LIN3060J
REMOTEREMOTE

GPIB

OTHER

HARD COPY

FILE

HOLD

EXT CONTSYSTEM

REMOTELOCAL

KEY LOCK
VIEW

START

F1
F2
USB

F3
F4

REMOTE

F5
F6

STOP

ENTRY
7

8

9

SHIFT

V RANGE
4

I RANGE
5

V RANGE
1I RANGE
23ESC

BUZZER
0

.

-

ENTER

AC/DC power supply

6

BUZZERBACKLIGHT

POWER

IEC dip
simulator
DSI series *1

PCR-WEA2
AC/DC power supply

PCR-WEA/WEA2

POWER

POWER

This system can simulate various conditions of phenomena occurringin AC power environments. It can be used for immunity tests of electrical and electronic devices, which are connected to a low-voltage distribution system, or which have DC power input ports, under the standard
conditions as specified to the right. The test conditions can be set
outside the standard range, allowing the system to be used for preliminary tests prior to standard tests, immunity-margin tests, and stresstests. The KHA3000 harmonic/flicker analyzer combines a PCR-WEA/
WEA2 Series AC power supply, LIN Series line-impedance network *1,
DSI series IEC dip simulator and application software(Refer to pg.8),
allowing tests that conform to IEC standards and JIS standards.
*1 Specially made to order

IEC61000-4-11

Voltage dipping, instantaneous power failure and voltage variation

IEC61000-4-13

Higher harmonics wave/interharmonic wave

IEC61000-4-14

Voltage swing

IEC61000-4-27

Unbalance in units

IEC61000-4-28

Variation in power-supply frequency for units with 16 A/phase

IEC61000-4-34

Voltage drop(dip), instantaneous power failure and voltage variation
for units with input current exceeding 16 A/phase

IEC61000-4-17

Ripple at the DC input power terminal

IEC61000-4-29

Voltage drop(dip), instantaneous power failure and voltage variationin DC *2

IEC61000-3-2,12Harmonic electric current limit level

IEC61000-3-3,11

Voltage fluctuation, Flicker limit level*2 Designed for preliminary test purposes.

For Testing of the EV Charging System● EV charging system (item under test)

Quick Charge

Residential Charge

DCHigh voltage
battery system

AC100 V/200 V

Power Line Disturbance
Immunity Testing SoftwareACcontroller

DC Power Supply
PXB Series

EV charging
system

DC400 V

AC/DC Power Supply
PCR-WEA Series

Low voltage
battery

IEC Dip Simulator
DSI Series

7Simple, user-friendly application software forvarious standard testing!
EMC

IEC61000-4Power Line Disturbance Immunity Testing Software

SD009-PCR-LE/WE [ Quick Immunity Sequencer 2 ]List of conformance to the EMC standard tests✔ : Conforming as standard▲ : Partially non-conforming

The latest standards for IEC61000-4 supported!

– : Function not available
Conforming
Single-phase Three-phase
✔ *1
✔ *1
✔ *1
✔ *1
✔
✔
✔
✔
✔
✔
✔
✔
✔
✔
✔
✔
✔
✔
✔
✔
✔
✔
✔
✔
✔
✔
–
✔
–
✔

Standard

Item

IEC61000-4-11

Ripple at the DC input power terminal

Voltage drop (dip)Instantaneous power failure
Voltage variation
Flat curve
Over swing
Frequency sweep
Odd harmonics the order of which is not a multiple of 3
Odd harmonics the order of which is a multiple of 3
Even harmonics
Interharmonics
Meister curve
Voltage swing
Interval
Single-phase rectifier circuit
Three-phase rectifier circuit

IEC61000-4-27 Unbalance in units
IEC61000-4-28

Unbalance–▲ *2

Frequency variation

✔

✔

✔
▲ *3
✔
▲ *4
▲ *4
✔

–
–
–
▲ *4
▲ *4
✔

Voltage dipping, instantaneous power failure
and voltage variation

IEC61000-4-13

Higher harmonics wave/interharmonic wave

IEC61000-4-14
Voltage swing

IEC61000-4-17

Variation in-power supply frequency for units with 16 A/phase

Voltage drop (dip)Instantaneous power failure
Voltage variation
Voltage drop (dip)IEC61000-4-34
Voltage drop (dip), instantaneous power failure and voltage Instantaneous power failure
variation for units with input current exceeding 16 A/phase Voltage variation

IEC61000-4-29

Voltage drop (dip), instantaneous power failure
and voltage variation in DC“Quick Immunity Sequencer 2” (model name: SD009-PCR-LE/WE) is an application software for immunity testing with the AC power supply PCR-WEA/WEA2 series system,
based on the power line disturbance standard (IEC61000-4 Series) for the immunity
testing of the EMC standard. Not only can it be used for compliance testing based on
the latest standards or for some types of preliminary testing, but the software can bealso employed for advance checking in development phases and for immunity margin
tests, because it allows extended testing conditions to be set as needed.

* Immunity testing for units with 16 A/phase except for those required by IEC61000-4-34*1 Conforms to the standard when used in combination with IEC Dip Simulator DSI series. If using the PCR-WEA/WEA2 alone, the voltage dips and short-time power failures are preliminary tests.
*2 110 %, 95.2 %, 93.5 %, 90 %, 87 %, 80 %, 74 %, 71 %, 66 % need to respond to sudden changes of 1 µs to 5 µs. The voltage response of PCR-WEA/WEA2 is more than 40 μs at FAST, which is a preliminary test.
*3 Must support output impedance greater than 100 kΩ. The PCR-WEA/WEA2 output impedance is less than 100 kΩ and therefore designed for preliminary testing purposes. Only PCR12000WEA2R conforms to the standards.
*4 The device between the range of 16A to 75 A requires having the capability of rapid change with 1 μs to 5 μs. The device exceeding 75 A is not required to have the capability of rapid change with 1μs to 5 μs.
(It is relaxed to 1 µs to 50 µs for the device exceeding 75 A.)

Avionics Test Software

SD012-PCR-LE/WE

Supported Standards
Military Standard:MIL-STD-704A/E/F
Civilian Standard:RTCA DO-160F/G
Civilian Standard:JIS W0812:2004

PCR-WEA Output Setting"MIL-STD-704" and "DO-160" tests
can be easily performed!
Avionics Test Software is software for power standards testingof onboard aircraft equipment. It corresponds to the defense
standard "MIL-STD-704", civilian standard "DO-160," and "JISW0812".

Standard Selection

Test Run

Output Monitor■ Easy to use

StatusSimply select the test item and choose the number you
wish to run from the test list.

■ Test items can be edited and saved

It is possible to create dedicated tests for development
evaluation and simple confirmation tests.

■ CSV output

Supports measurement data during testing and the output of test reports.

■ Supports 360 V DO-160 Abnormal Test

Surge Voltage 360 V testing of the DO-160 230 V AC
system is possible in combination with OT03-PCR-WEA.
* Separate PBZ20-20A series bipolar power supply is required when
performing the SD-160 Section 18 test.
* Please contact us for the MIL-STD-461 CS101 test.

8

Step List Editing Screen

Step List

Test Selection

Click Button
to Run Step

[Main Screen][Test Details (Step List) Editing Screen]

Defense Standards
Standard Name
MIL-STD-704A

Revision/
Category
Rev. A
Rev. E
Rev. F

SAC

Interchange Testing
SVF
TAC
TVF

SXF

DC Testing
LDC
HDC

✔

–

✔

–

–

✔

–

✔

–

✔

–

–

✔

✔

✔

✔

✔

✔

✔

✔

✔

Interchange Testing
A CF
A NF *3 A WF *3

ADC Testing
D
B

Z

✔

▲

▲

✔

✔

✔

✔

✔

▲

▲

✔

✔

✔

✔

✔

▲

▲

✔

✔

✔

✔

Civil Standards *1
Standard Name
RTCA/DO-160
Section 16
JIS W0812 *2

Revision/
Category
Rev. F
Rev. G
2004*1 Not applicable to primary standard test items• 16.5.1.5.1 Normal Surge Voltage and 16.5.2.3.1 Abnormal Surge Voltage for 230 V equipment at RTCA/DO-160F and RTCA/DO-160G• 16.5.1.5.1 Normal Surge Voltage and 16.5.3.3.1 Abnormal Surge Voltage for 230 V equipment in JIS W0812*2 Japanese domestic standard corresponding to DO-160 Rev. E*3 At a frequency of 600 Hz or higher, the voltage may exceed -1 % of the set value.

Wavy“Wavy” sequence creation software

SD032-PCR-WE [ Wavy for PCR-WE ]The software extends the feature of waveform generation andsequence functions.
Easy sequence control without programming knowledge!

Trial version
available on
website!
https://global.kikusui.
co.jp/downloads/

Download !
Wavy is an application software that supports
sequence creation and the operation for Kikusui
power supplies and electronic loads.
Wavy allows you to create and edit sequences
visually with a mouse without programming
knowledge.

▲Commandcontrol▲Direct control▲

Graph viewer/Configuration■ Makes it easier to create or edit the test-condition file required for the sequence operation.
■ By using the storage function of test-condition data file, it enables you to manage the test condition of the standard routine test.
■ The progress of execution sequence will be displayed in "practical dialogue" with the setting value and the cursor.
■ It is possible to observe the intuitionistic output through the "monitor graph" that plots the ongoing monitor value.
■ You can save the acquired monitor data as a test result.
■ Added "waveform image" window let's you easily keep track of the AC signal.
■ Allows you to edit and create a new arbitrary waveform easily. You can instantly write and then output the created arbitrary waveform.
■ You can select or deselect the pause function, trigger function, AC waveform etc. as necessary.

Remote-control software for Windows tablet

SD021-PCR-LE/WE [ RMT CONT SOFTWARE FOR PCR-LE/WE ]Windows tablet can be used as a remote controller !
The SD021-PCR-LE/WE is software that can control the PCR-WEA/
WEA2 Series. It is capable of changing the setting condition of the"wiring method", "output mode", "voltage range", "voltage value",and "frequency value". And the settings changed by remote control
can be saved and recalled. Moreover, it can display the measurement
value of the AC power supply. Remote operation and control of the
AC power supply can be easily achieved from a distance.
● Operating environment : Intel Core 2 or later / Windows 8.1 /Memory 4 GB / Storage 128 GB / Display resolution 133 x 768 or higher/ USB port

Screen lockCommunication

Voltage value

Wiring method

Frequency value

Output mode
Voltage range

OutputMonitor

Screen display (main screen)9Exterior Design
Front Panel
PCR1000WEA/2000WEA/3000WEA2

RS232C port

USB port

Control panel

USB memory connector*Not used for control.

POWER switch
Power ON/OFF switch

Power module air inlet
Contains an intake port and dust filter for internal cooling of the power unit.
*The eco-friendly function (energy-saving operation function) is not appliedwith PCR1000WEA, PCR2000WEA and PCR3000WEA2.

PCR12000WEA2

RS232C port

USB port

Control panel

USB memory connector*Not used for control.

Power module air inlet
Contains an intake port and dust
filter for internal cooling of thepower unit.
*The eco-friendly function(energy-saving operation function)
is applied with 6 kVA or more.
(See P5.)10POWER switch

POWER SELECTOR switch

Power ON/OFF switch

This switch is used to select between themaster unit and the slave unit operation
when multiple PCR-WEA units are used.

Rear Panel● PCR1000WEA/2000WEA/3000WEA2 *The photo shows the PCR3000WEA2. Covers have been removed for photographic purposes.
DIGITAL I/O connector

USB port (device)

Digital control connector.

USB port for remote control.

Power module air outlet

LAN port
LAN port for remote control.

ANALOG IN connector
Analog control connector.

AC INPUT terminal block

Expansion slot

Power inlet with a cover.

OUTPUTterminal block

SENSING terminal block
Connects the sensing cables.

Output terminal block with a cover.

● PCR6000WEA2R

Expansion slot

ANALOG IN connector
Analog control connector.

* Covers have been removed forphotographic purposes.

DIGITAL I/O connector
Digital control connector.

LAN port
LAN port for remote control.

PARALLEL
connector

USB port (device)
USB port for remote control.

Power module air outlet

SENSINGterminal block

AC INPUT terminal block(Three-phase three-wire)

Connects the sensing cables.

Models with an "R" in the name (6 kVA or
higher) have a three-phase, three-wire input.
For power cord connection (with cover)J2 connector

J1 connector

Power sync connector forparallel operation.

Power sync connector for parallel operation.

OUTPUT terminal block
Output terminal block with a cover.

● 400 V input model (Three-phase four-wire input)
Models of 6 kVA or more without an "R" in the name have three-phase
4-wire input.
*The photo shows the PCR36000WEA2. Covers have been removed for photographic purposes.

AC INPUT terminal block(Three-phase four-wire)
For power cord connection(with cover)●PCR1000WEA●PCR2000WEA●PCR3000WEA2●PCR6000WEA2●PCR6000WEA2R●PCR12000WEA2●PCR12000WEA2R●PCR18000WEA2●PCR18000WEA2R●PCR24000WEA2●PCR24000WEA2R●PCR30000WEA2●PCR30000WEA2R●PCR36000WEA2●PCR36000WEA2R

11

Unless specified otherwise, the specifications are for the following settings and conditions.
• The warm-up time is 30 minutes (with current flowing). • TYP: These are typical values that are representative of situations where the product operates
in an environment with an ambient temperature of 23°C. These values do not guarantee the performance of the PCR-WEA/WEA2.
• setting: Indicates a setting. • reading: Indicates the readout value. • f.s: Indicates full scale.

Specifications
Input (AC rms)

Single-phase output
Model

Nominalinput
voltage

1P2W input model

PCR
1000WEA

PCR
2000WEA

Single-phase/three-phase switchable model
PCR
3000WEA2

PCR
6000WEA2

PCR
12000WEA2

PCR
18000WEA2

PCR
24000WEA2

PCR
30000WEA2

PCR36000WEA2

PCR
6000WEA2R

PCR
12000WEA2R

PCR
18000WEA2R

PCR
24000WEA2R

PCR
30000WEA2R

PCR
36000WEA2R

100 Vac to 120 Vac / 200 Vac to 240 Vac *1―

3P3W 200 V input model―

200 Vac to 240 Vac (3 phase line voltage) *2

3P3W 400 V input model―

380 Vac to 480 Vac (3 phase line voltage) *2 *6

3P4W input model―

380 Vac to 480 Vac (3 phase line voltage) *3

Phase

Single-phase

Three-phase

Nominal input Frequency

50 Hz to 60 Hz

Input frequency range
Apparent power

45 Hz to 65 Hz
1.4 kVA and less 2.7 kVA and lessPower factor *4
1P2W input model *1

4 kVA and less0.95(TYP)
17 A / 8.5 A

32 A / 16 A

Maximum 3P3W 200 V input model
current
3P3W 400 V input model
3P4W input model

7.8 kVA and less 15.6 kVA and less 23.4 kVA and less 31.2 kVA and less 39 kVA and less 46.8 kVA and less
200 V: 0.97(TYP), 400 V: 0.95(TYP) 3P3W input model / 0.95(TYP) 3P4W input model

48 A / 24 A

―

―

27 A

53 A

80 A

106 A

133 A

―

14 A

28 A

42 A

56 A

70 A

84 A

―

14 A

28 A

42 A

56 A

70 A

84 A

Hold-up time for power interruption *4

159 A

10 ms

1P2W input model

3.5 mA or lessProtective 3P3W 200 V input model
conductor
current *5 3P3W 400 V input model
3P4W input model――

10 mA or less

15 mA or less

20 mA or less

25 mA or less

30 mA or less

35 mA or less―

10 mA or less

20 mA or less

30 mA or less

40 mA or less

50 mA or less

60 mA or less―

3.5 mA or less

*1 100 V/200 V input system (auto select) *2 Models with an "R" in the name (6 kVA or higher) have a three-phase, three-wire input. *3 Models of 6 kVA or more without an "R" in the namehave three-phase 4-wire input. *4 At output voltage 100 V/200 V, rated output current, sine wave, load power factor 1, output frequency 40 Hz to 1 kHz *5 At output voltage 100 V/200 V,
rated output current, sine wave, load power factor 1, output frequency 45 Hz to 65 Hz *6 For the 400V input model, do not connect to transformer equipment with isolation voltage exceeding 300 V. It cannot be connected to a delta-connected power distribution facility with one pole grounded.

OutputSingle-phase output
Model

PCR
1000WEA

PCR
2000WEA

Single-phase/three-phase switchable model
PCR
3000WEA2

Maximum peak current *11
Inrush current capacity *3

PCR
6000WEA2

PCR
12000WEA2

PCR
18000WEA2

PCR
24000WEA2

PCR
30000WEA2

PCR36000WEA2

PCR
6000WEA2R

PCR
12000WEA2R

PCR
18000WEA2R

PCR
24000WEA2R

PCR
30000WEA2R

PCR
36000WEA2R

4 times the maximum output current
3 times the rated current (0.07 s) *12
1.4 times the rated current (0.5 s)

1.4 times the rated current (0.5 s)

82 %(TYP)

85 %(TYP)

Efficiency *10
AC voltage
Rating160 V / 320 V *2

Setting range
AC
voltage*1

0 V to 161.0 V, 0 V to 322.0 V

Setting resolution

0.1 V

Setting accuracy(phase voltage) *3 *4±(0.3 % of setting + 0.3 V), ±(0.3 % of setting + 0.6 V)

Setting accuracy(Line voltage) *3 *4Maximum Single-phase output
current
Single-phase three-wire output,*1 *6
Three-phase output±(0.3 % of setting + 0.3 V), ±(0.3 % of setting + 0.6 V) *5
10 A / 5 A
―

Phase

30 A / 15 A

60 A / 30 A

120 A / 60 A

180 A / 90 A

240 A / 120 A

300 A / 150 A

360 A / 180 A

10 A / 5 A

20 A / 10 A

40 A / 20 A

60 A / 30 A

80 A / 40 A

100 A / 50 A

120 A / 60 A

3 kVA

6 kVA

12 kVA

18 kVA

24 kVA

30 kVA

36 kVA

2 kVA

4 kVA

8 kVA

12 kVA

16 kVA

20 kVA

24 kVA

1P
Single-phase output

Powercapacity

20 A / 10 A

1 kVA

1P2W, 1P3W, 3P4W switchable
2 kVA

Three-phase output―

Single-phase
three-wire output

Load power factor

0 to 1 (leading or lagging)

Setting range

1 Hz to 5 kHz *7 (5 kHz -3dB, <40 Hz derating required)

Frequency Resolution

0.01 Hz(1.00 Hz to 100.0 Hz), 0.1 Hz(100.0 Hz to 1000 Hz), 1 Hz(1000 Hz to 5000 Hz)

Accuracy *3
Phase±0.01 %, Temperature coefficient : ±0.005 %/°C

Resolution―

0.01*13 , 0.1° (1 Hz to 500 Hz), 1° (500 Hz to 4 kHz), 2° (4 kHz or more)

Accuracy *3―

Within ±(0.4° + fo×0.9°) *8 fo: frequency [kHz]DC voltage
Rating *1
DCvoltage-226 V to +226 V, -452 V to +452 V *2

Setting range *1-227.5 V to +227.5 V, -455.0 V to +455.0 V

Resolution

0.1 V

Accuracy *9±(0.05 % of setting +0.1 V)

Maximum current *6
Power capacity

10 A / 5 A

20 A / 10 A

30 A / 15 A

60 A / 30 A

120 A / 60 A

180 A / 90 A

240 A / 120 A

300 A / 150 A

360 A / 180 A

1 kW

2 kW

3 kW

6 kW

12 kW

18 kW

24 kW

30 kW

36 kW

*1 output L range, output H range *2 Specification guaranteed voltage range is 1 V to 160 V/ 2 V to 320 V (AC) and 1.4 V to 226 V/ 2.8 V to 452 V (DC) *3 At ambient temperature of 23°C±5 °C. *4 No load, output frequency 45 Hz to 65 Hz *5 When the phase angle of 120° of each phase. *6 For output phase voltage of 100 Vac to 160 Vac/ 200 Vac to 320 Vac and output voltage of 100 Vdc to 226 Vdc/ 200 Vdc to 452 Vdc, output current is reduced with output voltage. When the output frequency is between 1 Hz and 40 Hz, the output current is reducedby the output frequency. The output current is 70 % at 1 Hz. *7 On the 500 Hz limit model, the frequency is limited to 1 Hz to 500.0 Hz for three-phase output. *8 Within ±(0.4° + 2.5
μs×360°×fo×103). The following show the angles obtained by calculating the expression with the specified frequency, within ± 0.5° (at 60 Hz output), within ± 0.8° (at 400 Hz output)*9 With no load at 23°C±5°C. *10 When the output voltage is 100 V or 200 V, the output current is the rated value, the load power factor is 1, and the output frequency is between 40 Hz and
1 kHz. *11 Repeated output is possible when the crest factor is 4. *12 125 Vac/ 250 Vac (output L range/ H range) *13 Waveform bank 0, at 1 Hz to 500 Hz.

12Regeneration Function

Only for three-phase, three-wire input models with R at the end of the model name. Single-phase output models and three-phase, four-wire input models
do not have a regeneration function. For regeneration within the installation site only.
Single-phase/three-phase switchable model
Model

PCR
6000WEA2R

Maximum regenerated power *1
Maximum reverse
power flow current
*1 *2

PCR
12000WEA2R

PCR
18000WEA2R

PCR
24000WEA2R

PCR
30000WEA2R

PCR
36000WEA2R

6 kVA

12 kVA

18 kVA

24 kVA

30 kVA

36 kVA

1P2W

60 A / 30 A

120 A / 60 A

180 A / 90 A

240 A / 120 A

300 A / 150 A

360 A / 180 A

1P3W
3P

20 A / 10 A

40 A / 20 A

60 A / 30 A

80 A / 40 A

100 A / 50 A

120 A / 60 A

Regeneration efficiency *3

85 %(TYP)

Output current harmonic distortion

THD: 5 % and less, each harmonic: 3 % and less (2nd to 40th)*1 When the output phase voltage is between 100 Vac and 161 Vac or 200 Vac and 322 Vac, the output current is reduced by the output voltage.
When the output frequency is between 1 Hz and 40 Hz, the output current is reduced by the output frequency. The output current is 70 % at 1 Hz.
*2 When the output voltage is 100 V or 200 V and the output frequency is between 40 Hz and 1 kHz (when the current phase is -90 deg to -180 deg or 90 deg to 180 deg
relative to the output voltage)*3 When the output voltage is 100 V or 200 V, the output current is the rated value, sine wave, the load power factor is 1, and the output frequency is between 45 Hz to 65 Hz.

Output Voltage Stability (Phase Voltage)Single-phase output
Model

PCR
1000WEA

PCR
2000WEA

Single-phase/three-phase switchable model
PCR
3000WEA2

PCR
6000WEA2

PCR
12000WEA2

PCR
18000WEA2

PCR
24000WEA2

PCR
30000WEA2

PCR36000WEA2

PCR
6000WEA2R

PCR
12000WEA2R

PCR
18000WEA2R

PCR
24000WEA2R

PCR
30000WEA2R

PCR
36000WEA2RLine regulation *1

Within ±0.1 %Within ±0.1 V/ ±0.2 V(1 Hz to 100 Hz)
Within ±0.3 V/ ±0.6 V(100.1 Hz to 500 Hz)
Within ±1 V/ ±2 V(500.1 Hz to 1 kHz)

Load regulation *2

Within ±0.2 V/ ±0.4 V(1 Hz to 100 Hz)
Within ±0.3 V/ ±0.6 V(100.1 Hz to 500 Hz)
Within ±1 V/ ±2 V(500.1 Hz to 1 kHz)When the output voltage correction function is enabled : Within ±0.3 %(1 Hz to 1 kHz), Within ±10 %(1001 Hz to 5 kHz)When the output voltage correction function is disabled : Within -3 dB(5 kHz)

Output frequency variation *3
Ripple noise *4≤ 0.25 Vrms

Ambient temperature variation *5±100 ppm/ °C (TYP)

Total harmonic distortion *6

0.3 % and less(1 Hz to 100 Hz), 0.5 % and less(100.1 Hz to 330 Hz), 1.5 %/kHz and less(330.1 Hz to 5 kHz)

Transient response *7

Response FAST : 40 μs(TYP)

Response speed Tr/Tf *8

Response FAST : 40 μs(TYP)

Response MEDIUM : 100 μs(TYP)

Response SLOW : 300 μs(TYP)*1 With respect to changes in the rated range of input voltage.
*2 With respect to 0 % to 100 % changes in the rating of output current.
When the output phase voltage is between 80 V and 160 V (L range) or 160 V and 320 V (H range) and the load power factor is 1, and the response is FAST.
At the output terminal block, when the compensation function is not used.
*3 Voltage variation over 40 Hz to 5 kHz in AC mode with 55 Hz as the reference.
When the output phase voltage is between 80 V and 160 V or 160 V and 320 V and the load power factor is 1, and the response is FAST, at the output terminal block.
*4 5 Hz to 1 MHz components in DC mode.
*5 With respect to changes in the operating temperature range. When the output phase voltage is 100 V or 200 V, with no load.
*6 When the output phase voltage is between 80 V and 160 V or 160 V and 320 V and the load power factor is 1, and the response is FAST, at the output terminal block.
*7 When the output voltage is 100 V or 200 V, the load power factor is 1, and the output current changes from 0 A to the rated value and from the rated value to 0 A.
*8 At 10 % to 90 % of the output voltage.

Measurement
Single-phase output
Model

Resolution
Voltage
Rms value Accuracy *1
Resolution
Current
Rms value Accuracy *1 *2

PCR
1000WEA

PCR
2000WEA

Single-phase/three-phase switchable model
PCR
3000WEA2

PCR
12000WEA2

PCR
18000WEA2

PCR
24000WEA2

PCR
30000WEA2

PCR36000WEA2

PCR
6000WEA2R

PCR
12000WEA2R

PCR
18000WEA2R

PCR
24000WEA2R

PCR
30000WEA2R

PCR
36000WEA2R

0.1 V
DC, 40 Hz to 999.9 Hz : ±(0.3 % of reading +1 V), 1 kHz to 5 kHz : ±(0.5 % of reading +1 V)
0 to 99.99 A: 0.01 A, 100 to 999.9 A: 0.1 A
45 Hz to 65 Hz : ±(0.3 % of reading +0.3 % of f.s) DC, 40 Hz to 999.9 Hz : ±(0.6 % of reading +0.6 % of f.s)
1 kHz to 5 kHz : ±(1.2 % of reading +1.2 % of f.s)Resolution
Current
peak value Accuracy *1 *3
Activepower

PCR
6000WEA2

Resolution
Accuracy *1 *2 *4

0 to 99.99 A: 0.01 A, 100 to 999.9 A: 0.1 A
4 % of f.s
1 W *545 Hz to 65 Hz : ±(0.3 % of reading +0.3 % of f.s)Apparent power Resolution

1 VA *6

Power factor Resolution

0.01

Phase difference Resolution

0.1°

Frequency range(fundamental wave)

Harmonic
measure- Upper limit of harmonic analysisment
FFT data length
Measurement items
Recommended calibration period

10 Hz to 1 kHz
5th to 50th
4096
Rms voltage and current, phase angle, THD
1 year*1 At ambient temperature of 23 °C±5 °C.
*2 At 10 % to 100 % of maximum rated current, sine wave.
*3 Pulse height of sine wave
*4 At a power factor of 1.
*5 When the measured value is 0 to less than 100 W, the resolution is 0.1 W.
*6 When the measured value is 0 to less than 100 VA, the resolution is 0.1 VA.

13

Specifications
GeneralSingle-phase output
Model

PCR
1000WEA

PCR
2000WEA

Single-phase/three-phase switchable model
PCR
3000WEA2

PCR
6000WEA2

PCR
12000WEA2

PCR
18000WEA2

PCR
24000WEA2

PCR
30000WEA2

PCR36000WEA2

PCR
6000WEA2R

PCR
12000WEA2R

PCR
18000WEA2R

PCR
24000WEA2R

PCR
30000WEA2R

PCR
36000WEA2R

Between input and chasInsulationsis, output and chassis,
resistanceand input and output

500 Vdc, 10 MΩ or more

Between input and chasWithstandsis, output and chassis,
voltageand input and output

1500 Vac / 2150 Vdc, 1 minuteElectromagnetic compatibility(EMC) *1 *2

Complies with the requirements of the following directive andstandards.
EMC Directive 2014/30/EU
EN 61326-1 (Class A*3), EN 55011 (Class A*3, Group 1*4),
EN 61000-3-2*5, EN 61000-3-3*5
Applicable under the following conditions
The maximum length of all cabling and wiring connected to the
product must be less than 3 m.

Complies with the requirements of the following directive and standards.
EMC Directive 2014/30/EU
EN 61326-1 (Class A*3)
EN 55011 (Class A*3, Group 1*4)
Applicable under the following conditions
The maximum length of all cabling and wiring connectedto the product must be less than 3 m.

Complies with the requirements of the following directive and standards.
Low Voltage Directive 2014/35/EU*2 EN 61010-1 (Class I*6, Pollution Degree2*7)

Safety *1
Operating environment

Indoor use, overvoltage category II

Operating temperature range

0 °C to +50 °C (32 °F to +122 °F)

Environ- Storage temperature range
mentalconditions Operating humidity range
Storage humidity range

20 %rh to 80 %rh (no condensation)

Altitude

Up to 2000 m-10 °C to +60 °C (14 °F to +140 °F)
90 %rh and less (no condensation)

Dimensions

WeightSee page 17

Models without
regeneration function

18 kg(39.7 lb)

21 kg
(46.3 lb)

25 kg
(55.1 lb)

43 kg
(94.8 lb)

66 kg
(145.5 lb)

120 kg(264.6 lb)

130 kg(286.6 lb)

160 kg(352.7 lb)

180 kg(396.8 lb)

200 V input models withregeneration function

―

―

―

43 kg
(94.8 lb)

67 kg
(147.7 lb)

120 kg(264.6 lb)

130 kg(286.6 lb)

160 kg(352.7 lb)

180 kg(396.8 lb)

400 V input models withregeneration function

―

―

―

46 kg
(101.4 lb)

70 kg(154.3 lb)

120 kg(264.6 lb)

140 kg(308.6 lb)

170 kg(374.8 lb)

180 kg(396.8 lb)

Input terminal

M6

M5

Output terminal

M6

M5

3P3W input model: M8, 3P4W input model: M5
M6

M8

Cable tie (4 pcs.), External control(DIGITAL I/O) connector (1 pc.), Heavy object warning label (1 pc.)*Excludes PCR1000WEA,
Read This First! (1 copy), Quick Reference(1 sheet), CD-ROM (1 disc), Safety Information (1 copy)

Accessories*1 Does not apply to specially ordered or modified products.
*2 Only on models that have the CE marking on the panel.
*3 This is Class A equipment. This product is intended for use in an industrial environment. This product may cause interference if used in residential areas. Such use must be avoided
unless the user takes special measures to reduce electromagnetic emissions to prevent interference to the reception of radio and television broadcasts.
*4 This is Group 1 equipment. This product does not generate and/or use intentionally radio-frequency energy, in the form of electromagnetic radiation, inductive and/or capacitive coupling,
for the treatment of material or inspection/analysis purpose.
*5 This does not apply to the PCR6000WEA2R (3-phase 3-wire 200V input model).
*6 This is Class I equipment. Be sure to ground this product’s protective conductor terminal. The safety of this product is only guaranteed when the product is properly grounded.
*7 Pollution is addition of foreign matter (solid, liquid or gaseous) that may produce a reduction of dielectric strength or surface resistivity. Pollution Degree 2 assumes that only nonconductive pollution will occur except for an occasional temporary conductivity caused by condensation.

Output Impedance Setting
Resistance component
Single-phase output
Model

L rangeH range

1P
1P3W 3P
1P
1P3W 3P

PCR
1000WEA

PCR
2000WEA

0 Ω to 2000 mΩ 0 Ω to 1000 mΩ

Single-phase/three-phase switchable model
PCR
3000WEA2

PCR
12000WEA2

PCR
18000WEA2

PCR
24000WEA2

PCR
30000WEA2

PCR36000WEA2

PCR
6000WEA2R

PCR
12000WEA2R

PCR
18000WEA2R

PCR
24000WEA2R

PCR
30000WEA2R

PCR
36000WEA2R

0 Ω to 333 mΩ

0 Ω to 167 mΩ

0 Ω to 111 mΩ

0 Ω to 83 mΩ

0 Ω to 67 mΩ

0 Ω to 56 mΩ

0 Ω to 2000 mΩ 0 Ω to 1000 mΩ

0 Ω to 500 mΩ

0 Ω to 333 mΩ

0 Ω to 250 mΩ

0 Ω to 200 mΩ

0 Ω to 167 mΩ

0 Ω to 8000 mΩ 0 Ω to 4000 mΩ 0 Ω to 2667 mΩ 0 Ω to 1333 mΩ

0 Ω to 667 mΩ

0 Ω to 444 mΩ

0 Ω to 333 mΩ

0 Ω to 267 mΩ

0 Ω to 222 mΩ

0 Ω to 8000 mΩ 0 Ω to 4000 mΩ 0 Ω to 2000 mΩ 0 Ω to 1333 mΩ 0 Ω to 1000 mΩ

0 Ω to 800 mΩ

0 Ω to 667 mΩ

―
―

―
―

0 Ω to 667 mΩ

PCR
6000WEA2

Reactance component■ Response: FAST

Single-phase output
Model

1P
L range

1P3W
3P
1P

H range

14

1P3W
3P

Single-phase/three-phase switchable model
PCR
6000WEA2

PCR
12000WEA2

PCR
18000WEA2

PCR
24000WEA2

PCR
30000WEA2

PCR36000WEA2

PCR
6000WEA2R

PCR
12000WEA2R

PCR
18000WEA2R

PCR
24000WEA2R

PCR
30000WEA2R

PCR
36000WEA2R

PCR
1000WEA

PCR
2000WEA

PCR
3000WEA2

40 µH to2000 µH

20 µH to1000 µH

13 µH to667 µH

7 µH to333 µH

3 µH to167 µH

2 µH to111 µH

2 µH to
83 µH

1 µH to67 µH

1 µH to
56 µH

―

―

40 µH to2000 µH

20 µH to1000 µH

10 µH to500 µH

7 µH to333 µH

5 µH to250 µH

4 µH to200 µH

3 µH to167 µH160 µH to
8000 µH

80 µH to4000 µH

53 µH to2667 µH

27 µH to1333 µH

13 µH to667 µH

9 µH to444 µH

7 µH to333 µH

5 µH to267 µH

4 µH to222 µH――160 µH to
8000 µH

80 µH to4000 µH

40 µH to2000 µH

27 µH to1333 µH

20 µH to1000 µH

16 µH to800 µH

13 µH to667 µH■ Response: MED
Single-phase output
Model

1P
L range

1P3W
3P
1P

H range

1P3W
3P

Single-phase/three-phase switchable model
PCR
6000WEA2

PCR
12000WEA2

PCR
18000WEA2

PCR
24000WEA2

PCR
30000WEA2

PCR36000WEA2

PCR
6000WEA2R

PCR
12000WEA2R

PCR
18000WEA2R

PCR
24000WEA2R

PCR
30000WEA2R

PCR
36000WEA2R

PCR
1000WEA

PCR
2000WEA

PCR
3000WEA2

80 µH to2000 µH

40 µH to1000 µH

27 µH to667 µH

13 µH to333 µH

7 µH to167 µH

4 µH to111 µH

3 µH to
83 µH

3 µH to67 µH

2 µH to
56 µH

―

―

80 µH to2000 µH

40 µH to1000 µH

20 µH to500 µH

13 µH to333 µH

10 µH to250 µH

8 µH to200 µH

7 µH to167 µH320 µH to
8000 µH

160 µH to4000 µH

107 µH to2667 µH

53 µH to1333 µH

27 µH to667 µH

18 µH to444 µH

13 µH to333 µH

11 µH to267 µH

9 µH to222 µH――320 µH to
8000 µH

160 µH to4000 µH

80 µH to2000 µH

53 µH to1333 µH

40 µH to1000 µH

32 µH to800 µH

27 µH to667 µH■ Response: SLOW
Single-phase output
Model

1P
L range

1P3W
3P
1P

H range

1P3W
3P

Single-phase/three-phase switchable model
PCR
6000WEA2

PCR
12000WEA2

PCR
18000WEA2

PCR
24000WEA2

PCR
30000WEA2

PCR36000WEA2

PCR
6000WEA2R

PCR
12000WEA2R

PCR
18000WEA2R

PCR
24000WEA2R

PCR
30000WEA2R

PCR
36000WEA2R

PCR
1000WEA

PCR
2000WEA

PCR
3000WEA2

240 µH to2000 µH

120 µH to1000 µH

80 µH to667 µH

40 µH to333 µH

20 µH to167 µH

13 µH to111 µH

10 µH to
83 µH

8 µH to67 µH

7 µH to
56 µH

―

―

240 µH to2000 µH

120 µH to1000 µH

60 µH to500 µH

40 µH to333 µH

30 µH to250 µH

24 µH to200 µH

20 µH to167 µH960 µH to
8000 µH

480 µH to4000 µH

320 µH to2667 µH

160 µH to1333 µH

80 µH to667 µH

53 µH to444 µH

40 µH to333 µH

32 µH to267 µH

27 µH to222 µH――960 µH to
8000 µH

480 µH to4000 µH

240 µH to2000 µH

160 µH to1333 µH

120 µH to1000 µH

96 µH to800 µH

80 µH to667 µH

Limit Values and Protection Functions (Common Specification)
AC voltage upper limit
AC voltage lower limit

Voltageprotection

DC voltage upper limit
DC voltage lower limit
Outputovervoltage
protection(OVP)

Current
protection
Overheatprotection

Setting resolution

0.0 V to 322.0 V

0.1 V-455 V to 455 V

0.1 V

Rms value

14.0 V to 500.5 V

0.1 V

Positive peak value
Negative peak value

14.0 V to 500.5 V-500.5 V to -14.0 V

0.1 V

Fixed

―

Power module overvoltage protection

Frequency
protection

Setting range

Output undervoltage protection (UVP)

0.0 V to 500.5 V

0.1 V

Frequency upper limit
Frequency lower limit

1 Hz to 5000 Hz
500 Hz LMT model: 1 Hz to 500 Hz(Three-phase output)

0.01 Hz (1.00 Hz to 100.0 Hz)
0.1 Hz (100.0 Hz to 1000 Hz),
1 Hz (1000 Hz to 5000 Hz)

Current limit *1

Maximum output current × 0.1 to maximum
output current × 1.1

Positive peak current limit
Negative peak current limit *2

Maximum output current × 0.1 to maximum
output current × 4.2

0.01 A (0.35 A to 100.0 A),
0.1 A (100.0 A to 1000 A)

Power module overheat protection

Fixed

―

Fan error

Fixed

―

Overload protection

Rated current or current limit

Current limit resolution

Independent operation detection

Fixed

―

Sensing error detection±(10 % +10 V) with respect to the output
terminal voltage―*1 The current that can actually be supplied is 1.1 times the rated current or the current limit, whichever is less.
*2 The current that can actually be supplied is the maximum peak current or the current limit, whichever is less.

Communication Interface (Common Specification)
USB

Complies with the USB 2.0 specifications; data rate: 480 Mbps (high speed), socket B type, self-powered,Complies with the USBTMC-USB488 device class specifications.

LAN

IEEE802.3, 100Base-TX Ethernet
LXI Rev.1.5 2016 (extended functions: VXI-11, HiSLIP, IPv6), data rate: 100 Mbps (auto negotiation, full speed) AUTO MDIX function
IPv4, RJ45 connector, category 5, straight cableComplies with SCPI Specification 1999.0

RS232C

Complies with the EIA232D specifications, asynchronous full duplex, D-SUB 9-pin connector (male), crossover cable (null modem),
9600bps/19200bps/38400bps/57600bps/115200bps

GPIB (option)

Complies with IEEE Std 488.1-1987
SH1, AH1, T8, L4, SR0, RL0, PP0, DC0, DT0, C0, E1 24-pin connector (receptacle)15262 mm(10.32 inch)

3 times

the power*of previous PCR-W series

Actualsize
16

Dimensions (Unit:mm(inches))

MAX440(17.32)
430(16.93)• This figure shows 200 V model.
• The 400 V model includes aterminal block cover.

MAX20(0.79)

674(26.54)
MAX735(28.94)

563(22.17)
MAX670(26.38)

Terminal
block
cover

430(16.93)■PCR6000WEA2/
PCR6000WEA2R

690(27.17)

MAX785(30.91)

MAX150(5.91)
129.2(5.09)

MAX20(0.79)

430(16.93)■ PCR18000WEA2/ PCR18000WEA2R
PCR24000WEA2/ PCR24000WEA2R

MAX445(17.52)■PCR1000WEA/ PCR2000WEA/ PCR3000WEA2• This figure shows 200 V model.
• The 400 V model includes aterminal block cover.

430(16.93)■ PCR30000WEA2/ PCR30000WEA2R
PCR36000WEA2/ PCR36000WEA2R

MAX445(17.52)

563(22.17)
MAX635(25.00)

MAX345(13.58)
262(10.32)

MAX20(0.79)

MAX20(0.79)

563(22.17)
MAX670(26.38)

944(37.17)563(22.17)
MAX635(25.00)

MAX475(18.70)
389(15.32)

MAX20(0.79)

MAX1040(40.94)430(16.93)■PCR12000WEA2/
PCR12000WEA2R

Terminal
block
cover

● Dimensions
Dimensions(mm(inch))(Maximum size)

Model

Dimensions(mm(inch))(Maximum size)

PCR1000WEA

Model

430(16.93")(440(17.32"))W×129.2(5.09")(150(5.91"))H×674(26.54")(735(28.94"))Dmm

PCR18000WEA2R

430(16.93")(445(17.52"))W×690(27.17")(785(30.91"))H×563(22.17")(670(26.38"))Dmm

PCR2000WEA

430(16.93")(440(17.32"))W×129.2(5.09")(150(5.91"))H×674(26.54")(735(28.94"))Dmm

PCR18000WEA2

430(16.93")(445(17.52"))W×690(27.17")(785(30.91"))H×563(22.17")(670(26.38"))Dmm

PCR3000WEA2

430(16.93")(440(17.32"))W×129.2(5.09")(150(5.91"))H×674(26.54")(735(28.94"))Dmm

PCR24000WEA2R

430(16.93")(445(17.52"))W×690(27.17")(785(30.91"))H×563(22.17")(670(26.38"))Dmm

PCR6000WEA2R

430(16.93")W×262(10.32")(345(13.58"))H×563(22.17")(635(25.00"))Dmm

PCR24000WEA2

430(16.93")(445(17.52"))W×690(27.17")(785(30.91"))H×563(22.17")(670(26.38"))Dmm

PCR6000WEA2

430(16.93")W×262(10.32")(345(13.58"))H×563(22.17")(635(25.00"))Dmm

PCR30000WEA2R

430(16.93")(445(17.52"))W×944(37.17")(1040(40.94"))H×563(22.17")(670(26.38"))Dmm

PCR12000WEA2R

430(16.93")W×389(15.32")(475(18.70"))H×563(22.17")(635(25.00"))Dmm

PCR30000WEA2

430(16.93")(445(17.52"))W×944(37.17")(1040(40.94"))H×563(22.17")(670(26.38"))Dmm

PCR12000WEA2

430(16.93")W×389(15.32")(475(18.70"))H×563(22.17")(635(25.00"))Dmm

PCR36000WEA2R

430(16.93")(445(17.52"))W×944(37.17")(1040(40.94"))H×563(22.17")(670(26.38"))Dmm

PCR36000WEA2

430(16.93")(445(17.52"))W×944(37.17")(1040(40.94"))H×563(22.17")(670(26.38"))Dmm

17

Options■ GPIB Interface Boards
IB07-PCR-WE■ External-control Connector
OP01-PCR-WE (for DIGITAL I/O)

This board enables you to control the PCR-WEA/
WEA2 Series over GPIB.

■ External-control Connector
OP02-PCR-WE (for ANALOG I/O)

■ Parallel-operation Cable (1 m)
PC01-PCR-WE■ Rack Mount Brackets
For PCR1000WEA/2000WEA/3000WEA2
KRB3-TOS (EIA inch rack)
KRB150-TOS (JIS millimeter rack)■ Power-sync Cable (1 m)
LC01-PCR-LE

For PCR6000WEA2(R)KRB6 (EIA inch rack)
KRB300 (JIS millimeter rack)■ Base Hold Angles
OP03-KRCFor PCR12000WEA2(R)KRB9 (EIA inch rack)
KRB400-PCR-LE (JIS millimeter rack)■ Input Power Cable
Appropriate Model

Model

Cable

LengthNominal crosssectional area

Input
terminal

PCR1000WEA/2000WEA

1P2W input

AC5.5-1P3M-M6C-3S

Single core, 3 pcs.

3m

5.5 mm2

M6

PCR3000WEA2

1P2W input

AC14-1P3M-M6C-3S

Single core, 3 pcs.

3m

14 mm2

M6

PCR6000WEA2R

3P3W 200 V input

AC5.5-1P3M-M5C-4S

Single core, 4 pcs.

3m

5.5 mm2

M5

PCR6000WEA2R

3P3W 400 V input

AC5.5-1P3M-M5C-4S

Single core, 4 pcs.

3m

5.5 mm2

M5

PCR6000WEA2

3P4W 400 V input

AC5.5-1P3M-M5C-5S

Single core, 5 pcs.

3m

5.5 mm2

M5

PCR12000WEA2R

3P3W 200 V input

AC14-1P3M-M5C-4S

Single core, 4 pcs.

3m

14 mm2

M5

PCR12000WEA2R

3P3W 400 V input

AC5.5-1P3M-M5C-4S

Single core, 4 pcs.

3m

5.5 mm2

M5

PCR12000WEA2

3P4W 400 V input

AC5.5-1P3M-M5C-5S

Single core, 5 pcs.

3m

5.5 mm2

M5

PCR18000WEA2R

3P3W 200 V input

AC22-1P3M-M8C-4S

Single core, 4 pcs.

3m

22 mm2

M8

PCR18000WEA2R

3P3W 400 V input

AC8-1P3M-M8C-4S

Single core, 4 pcs.

3m

8 mm2

M8

PCR18000WEA2

3P4W 400 V input

AC8-1P3M-M5C-5S

Single core, 5 pcs.

3m

8 mm2

M5

PCR24000WEA2R

3P3W 200 V input

AC38-1P3M-M8C-4S

Single core, 4 pcs.

3m

38 mm2

M8

PCR24000WEA2R

3P3W 400 V input

AC14-1P3M-M8C-4S

Single core, 4 pcs.

3m

14 mm2

M8

PCR24000WEA2

3P4W 400 V input

AC14-1P3M-M5C-5S

Single core, 5 pcs.

3m

14 mm2

M5

PCR30000WEA2R

3P3W 200 V input

AC60-1P3M-M8C-4S

Single core, 4 pcs.

3m

60 mm2

M8

PCR30000WEA2R

3P3W 400 V input

AC22-1P3M-M8C-4S

Single core, 4 pcs.

3m

22 mm2

M8

PCR30000WEA2

3P4W 400 V input

AC22-1P3M-M5C-5S

Single core, 5 pcs.

3m

22 mm2

M5

PCR36000WEA2R

3P3W 200 V input

AC60-1P3M-M8C-4S

Single core, 4 pcs.

3m

60 mm2

M8

PCR36000WEA2R

3P3W 400 V input

AC22-1P3M-M8C-4S

Single core, 4 pcs.

3m

22 mm2

M8

PCR36000WEA2

3P4W 400 V input

AC22-1P3M-M5C-5S

Single core, 5 pcs.

3m

22 mm2

M5■ Sequence Creation Software "Wavy"
SD032-PCR-WE (Wavy for PCR-WE)
The software that further enhances the waveform generation andsequence functions of the PCR-WEA/WEA2 Series.
Easy sequence control without programming knowledge!

*Screen sample

18

Wavy is an application software that supports sequence creation and operation for Kikusui power
supplies and electronic loads. Wavy allows you to create and edit sequences visually with just amouse. Real-time graph-monitor function is equipped and enables monitoring and logging values ofvoltage and current. It is possible to operate the power supply with the feeling of remote control bydirect control function.

■ Output Terminal Box

Easy to select output mode
“single-phase, single-phase 3-wire, and 3-phase” without re-wiring.
● 2 lineups depend on output power, “6 kVA to 18 kVA model” and “24 kVA to 36 kVA model”.
● Toggle between “single-phase” or “single-phase 3-wire/3-phase” output terminal using main unit switch.
An output terminal box gives output
mode selection “single-phase, single-phase 3-wire and 3-phase” of
PCR-WEA/WEA2 series.
Selectable switches equipped in its
body achieve multi-phase output
without output cable re-wiring.

Changeover switch

Input terminal surface

OUT
single-phase 3-wire/
3-phase output

Output terminal surface■ Lineup

PCR-WEA sideINsingle-phase output

Connection concept diagram■ Connecting cable
Model

Model

Output terminal box (18 kVA)

OT01-PCR-WE

For 6 k, 12 kVA (0.7 m)

Output terminal box (36 kVA)

OT02-PCR-WE

Model

AC14-7P0.7M-M5M6

For 24 kVA (0.7 m)

AC22-7P0.7M-M6M8

For 6 k, 12 kVA (1.4 m)

AC14-7P1.4M-M5M6

For 24 kVA (1.4 m)

AC22-7P1.4M-M6M8

For 18 kVA (0.7 m)

AC22-7P0.7M-M6M6

For 30 k, 36 kVA (0.7 m)

AC38-7P0.7M-M8M8

For 18 kVA (1.4 m)

AC22-7P1.4M-M6M6

For 30 k, 36 kVA (1.4 m)

AC38-7P1.4M-M8M8■ Specification
Item
Connectable AC power supplies

OT01-PCR-WE

OT02-PCR-WE

PCR6000WEA2(R), PCR12000WEA2(R), PCR18000WEA2(R)PCR24000WEA2(R), PCR30000WEA2(R), PCR36000WEA2(R)

Maximum voltage (phase voltage)
Input/Output maximum Maximum current (Single-phase 2 wire)
rating (AC)Maximum current (Single-phase 3 wire/3-phase)

320 Vac
180 Aac360 Aac

60 Aac120 AacFrequency
Input terminal
Output terminal(Single-phase 2 wire)

45 Hz to 400 Hz

Type

M6 × 7P screw terminals

Arrangement/Quantity
Type/Arrangement/Quantity

Type
Output terminal(Single-phase 3 wire/3-phase) Arrangement/Quantity
Dimensions(W×H×D)/WeightM8 × 7P screw terminals
U-V-W-N-N-N-G /1 piece

M10 × 3P screw terminals/ L-N-G / 1 pieceM6 × 5P screw terminals

M8 × 5P screw terminals
U-V-W-N-G /1 piece

445 mm×215 mm×410 mm / Approx. 13 kg (28.7 lb)

445 mm×270 mm×410 mm / Approx. 19 kg (41.9 lb)■ 6 kVA single-phase/three-phase output transformer OT03-PCR-WEA

NEW

DO-160 AC 230 V system Surge Voltage 360 V test is possible!
The OT03-PCR-WEA high-voltage transformer is a step-up transformer that can be used with the PCR-WEA series andthe SD012-PCR-LE/WE avionics test software. Depending on the wiring, a single unit with 4U and 6 kVA output can beused in a single-phase or three-phase configuration.
● The device is designed to perform high-voltage tests onavionics equipment. Thanks to the built-in step-up transformer, it can handle 360 V DO-160 abnormal tests.
● Maximum output voltage: 440 Vrms
● Input voltage range: 100 V to 320 Vrms
● Voltage conversion ratio: input [1], output [1.34]● Frequency range: 45 Hz to 1200 Hz● THRU mode: High-voltage tests and routine tests can be
performed without reconnection.
● The standard number of units in parallel: 2 units (12 kVA)
* Please consult us if you would like to use more than two units.

■ OT03-PCR-WEA● Software control and the digital I/O ports used in the
PCR-WEA series can be used without adjustments.
● Voltage sensing function: In addition to the main unit's
sensing terminal, the voltage compensation function of
the SD012 enables stable 360 V output even in the 800
Hz band.

Input power test system for onboard
aircraft equipment
The input power supply test system for onboard aircraft equipment is a test system consisting of the AC power supply PCRWEA series, the high-voltage transformer OT03-PCR-WEA,
and the SD012-PCR-LE/WE avionics test software. The SD012PCR-LE/WE can be used for various power input tests required
by DO-160Section 16. Also, since the SD012-PCR-LE/WE includes the defense standard MIL-STD-704E/F, it can be used totest both commercial and defense aircraft components. It offersa highly versatile, all-in-one testing environment.
● Maximum capacity: AC 12 kVA / DC 12 kW● Compatible voltages: AC 115 V series / AC 230 V series● Wiring method: Switchable between single-phase andthree-phase● Testable frequencies:
DC / 400 Hz / 360 Hz to 650 Hz / 360 Hz to 800 Hz* To use OT03-PCR-WEA, SD012-PCR-LE/WE Ver. 2.00 orlater is required.
* When conducting the DO-160 Section 18 test, a separate
bipolar power supply PBZ20-20A series is required.
* Please contact us for the MIL-STD-461 CS101 test.

19

The never-ending evolutionof power supplies!
PCR-WEA/WEA2 Series
1

POWER

UP

points

Output voltage up 310 Vrms
►►►

4

2

320 Vrms

Transient response response speed 55 μs►►►

Improved output stability.
►►►

3

40 μs

Increased stability in
SLOW mode.

5

Output impedance reduction►►►

Reducedby 50 %

Interface options►►►

Analog monitor output● Comparison with previous model
Model

PCR-WE/WE2 Series

PCR-WEA/WEA2 Series

Firmware

Ver 1.24

Ver 3.12 or later

Basic function

Output voltage
155/310 Vrms
±219/438 Vdc

Output voltage
160/320 Vrms
±226/452 Vdc

Applied functions

Same value regardless of the lower limit
response setting for output impedance(reactance component).

Lower limit of output impedance (reactance component)
FAST : reduced by 50 %MED : no change
SLOW : 3x

Interface

None

Addition of analog monitor output option (factory option)* 6 kVA models and higher●Distributor:

1-1-3, Higashiyamata, Tsuzuki-ku, Yokohama, Kanagawa, 224-0023, Japan
Phone:(+81)45-593-0200, Facsimile:(+81)45-593-7591, https://global.kikusui.co.jp/■ All products contained in this catalogue are equipment and devices that are premised on use under thesupervision of qualified personnel, and are not designed or produced for home-use or use by general
consumers. ■ Specifications, design and so forth are subject to change without prior notice to improve thequality. ■ Product names and prices are subject to change and production may be discontinued whennecessary. ■ Product names, company names and brand names contained in this catalogue represent therespective registered trade name or trade mark. ■ Colors, textures and so forth of photographs shown in thiscatalogue may differ from actual products due to a limited fidelity in printing. ■ Although every effort has been
made to provide the information as accurate as possible for this catalogue, certain details have unavoidably
been omitted due to limitations of space. ■ If you find any misprints or errors in this catalogue, it would beappreciated if you would inform us. ■ Please contact our distributors to confirm specifications, prices,
accessories or anything that may be unclear when placing an order or concluding a purchasing agreement.

Printed in Japan

Issue: May 2023

202305PDFEC51