０．OUTLINE

1. Project outline and procedures
2. Equipment installation and acceptance3. Training content (text explanation)(1) Explanation of electrolyzed water(2) How to use the device(3) Hygiene management
4. monitoring(1) Operation of Aquapurita(2) Hygiene management status (cleaning test)

1１．Project outline and procedures: Project outline• In November 2020, the United Nations Industrial Development
Organization (UNIDO) adopted 13 Japanese companies with
useful technology projects to support for controlling newcoronavirus infections (COVID-19) and various infectious
diseases in developing countries• This project supports Japanese companies to disseminate
technologies for improving the health and hygiene environment ina total of 12 countries in Asia and Africa.

Source: ©AGC Inc.

１． Project outline and procedures: Procedures
The University of Danang (UD)
Needs confirmation, prepare
documents, agent to receive
equipment, contact local
government organizations…

confirmation

Hoa Vang hospital(A special hospitalfor Copronapatients)

Application
DanangDepartmentof Health

AnswerPermission
Contact / consultation / answer
Mailing / receiving equipment

Consultation on application
documents, permission
confirmationSubmitPermission
Danang city
Government

OEC, ACG companies,Japan

UNIDO, Japan

3

２．Equipment installation and acceptance (08 Feb, 2021)

Preparation of drain
Online installation

Extra pumpPrepare for operation

4２．Equipment installation and acceptance (Jan –Feb 2021)

After installed the equipment, we check each equipment to confirm their operation normally or not.

Equipment No
Test items

Standard

Production
capacity

1.28－1.90

Effective chlorine
concentration

>80

096

097

099

100

OKOK

OK

OK

98

91

97

107

5.5

5.0

6.0

6.0

OKOKOKOK㏗

2.7-6.5

Softening ofsupplying water<1mg/L

Electrolytic voltage

<10.5

4.6

4.7

4.3

4.9

Electrolytic

9.5

8.8

9.0

8.7

9.5

Water leakNone

OKOKOKOK

OtherNoneOKOKOK

OK5

３．Training(1)Explanation of electrolyzed water① Materials used
② Unit configuration③ Acidic electrolyzed water④ Bactericidal effect(2) How to use the equipment① How to operate② Inspection③ Precautions（３） Hygiene management① Bactericidal effect and cleaning degree② Cleaning degree management (cleaning degree
management by ATP inspection).
③ Implementation of live bacteria test

6３.Training (25 Feb, 2021)

Water softening check

Operation confirmation

effective chlorine concentration test

Check ㏗7

Staff training for testing and checking
Textbook for staff training
Explanation of electrolyzed water (sterilizationeffect, etc.)

Explanation of electrolyzed water generator

Explanation for obtaining an appropriate
sterilization effectExplanation of hygiene management by ATP
inspection

8

４．Monitoring (Equipment)（１）Aquapulita operation status

Equipment No
Test items

Standard096

097

099

100Production
capacity

1.28－1.90◯/◯◯/◯◯/◯◯/◯◯/◯◯/◯◯/◯◯/◯

Effective chlorineconcentration>80㏗

2.7-6.5◯/◯◯/◯◯/◯◯/◯

Softening ofsupplying water<1mg/L◯/◯◯/◯◯/◯◯/◯

Electrolytic voltage<10.5◯/◯◯/◯◯/◯◯/◯

Electrolytic9.5

◯/◯

◯/◯

◯/◯

◯/◯

Water leakNone◯/◯

◯/◯

◯/◯

◯/◯

OtherNone◯/◯◯/◯◯/◯◯/◯
9

Hoa Vang hospital is using the equipment effectively!

10