Quality color measurement within your grasp
CC-m is a Portable Spectrocolorimeter.
It is great for measuring the surface of large specimenson site. Its vertical structure allows the user to use it innearly any situation, such as measuring in narrow
spaces and measuring rough surfaces. Its large color
LCD touch panel makes it easy to read, prevents
mistakes, and reduces fatigue. The light source is
SUGA's distinctive VI-LED (High color-rendering index
white LED) with long lifespan and light suitable forcolor measurement. Its dual synchro sensor methodwith two spectroscopes gives it an outstanding
long-term stability.

is ibilit y・Easy to read・Prevents mistakes
・Reduces fatigue

pe r a bilit y・Simple to use
・Reduces operator error
・Intuitivee lia bilit y・Stable long term measurements
・VI-LED 20,000,000 flashes
・Stability less than 0.04 ⊿E*ab

Easy to see large-screen color LCD
The icons are large and visible with a 4.3 inchlarge-screen color LCD. The measurement
results are displayed on a large graph making
them easily visible for all users.

Select from:CIE LAB, CIE LAB color difference,
spectral reflectance, CIE Yxy, Lab(h),Lab(h) color difference

Operability with special attention tousability
All the necessary functions can be found at themain menu, allowing the user to operate withminimal amount of button-pressing. The M
button on the side of the device allows the user
to perform the three most commonly usedfeatures (zero standardization, full scalestandardization, and measurement).

All navigation begins at this single
screen.

Stability greatest of its class
The dual synchro sensor method simultaneously observes the reflective light from thesample and the lamp’ s light energy, wavelength
by wavelength. The light source is SUGA’ sTHE
DUAL SYNCHROSENSOR METHOD

VI-LED(High color-renderingindex white LED)distinctive VI-LED (high color-rendering indexwhite LED) with a lifespan of up to 20,000 hours(20 million measurements), which produces light
all across the visible spectrum, as opposed tothe traditional white LED. These features andsuperb reliable electronics makes the instrument
perform with outstanding long term stability.

LONG-TERM STABILITY

Standard deviation of ⊿E*ab within 0.04.

Measurement Systems
CIE LAB, CIE LUV, Spectral reflectance, CIE XYZ, CIE Yxy, Hunter LAB, Munsell values, JIS colorfastness
grade, whiteness index, yellowness index, change in yellowness index, pass/fail, color difference graphs.

Specifications
Measurement conditions

A, C, D₅₀, D₆₅, D₇₅, F₂, F₆, F₇, F₈, F₁₀, F₁₁, TL₈₄, UL₃₀
(2° and 10° observer angles)

Geometry

Diffuse light source, 8° observer angle (switchable between de:8° anddi:8°), reflectance measurement

Measurement aperture diameters

10 mm, 5 mm

Photometry

Dual synchro sensor method

Wavelength range

400 – 700 nm, 10 nm intervalLight source

VI-LED (High color-rendering index white LED)

Stability

The standard deviation of ⊿E*ab within 0.04
(measuring a white standard 30 times consecutively)

Dimension

Approx. 82(W) × 112(D) × 248(H) mm [weight: 980g]

Power capacity

3.7V rechargeable lithium-ion battery

Charging time

Approx. 7 hours (capable of approx. 1000 measurements at full charge)

Battery charger

AC 100 – 250V 50/60Hz 0.5A

Interfaces

USB2.0 miniB

Data storage

100 standard data, 1000 samples

Statistics

Average, standard deviation

Standards

JIS Z 8722, JIS L 0809, CIE Pub. No15, ISO 7724, ISO 11664,
ISO 105 J01 – 03, ASTM E313, ASTM E308, ASTM E1164

Languages

Japanese

EnglishGermanFrenchItalian

SpanishChineseKorean※1 The standard white calibration plaque, used as the standard for measurement values, is traceable to the international standard by National

Institute of Advanced Industrial Science and Technology. Accuracy of the instrument is maintained via our JCSS optical calibration technology.
Due to product development and improvement, specifications may be subject to change without noticeSeptember. 2014