STB-5037533350-1515-L1-C

Advanced

Description:
Model STB-5037533350-1515-L1-C is a compat broadband
benchtop low noise amplifier with a typical small signal gain of
33 dB and a nominal noise figure of 5 dB across the frequency
range of 50 to 75 GHz. The power supply required is a single
phase AC voltage in the range of 100 to 240 VAC, which can besupplied by a wall outlet. The unit is light weight, compact anddesigned for the purpose of easy biasing and reconfiguration for
applications requiring waveguide interface. The input andoutput port configurations are both WR-15 waveguide with UG-385/U flanges.

Features:•••

Applications:•••Full Waveguide Band CoverageState-of-the-Art Noise Figure
High Gain

IEEE 802.11.ad WiGigLow Noise Receivers
Lab Test Equipment

Electrical Specifications:Parameter
Frequency
Gain
Noise Figure
P1dB*
Maximum Input RF Power
Input Return Loss
Output Return Loss
Power Supply (AC Adapter Provided)
Specification Temperature
Operating Temperature

Minimum
50 GHzTypicalMaximum
75 GHzSAGE Millimeter, Inc.

Compact Bench Top Low Noise Amplifier, 50 to 75 GHz, 33 dB Gain, 5 dB NF33 dB
5 dB+8 dBm-20 dBm
7 dB
7 dB
100 VAC

240 VAC+25 ˚C

0 ˚C+50 ˚C* See Notes

Mechanical Specifications:
ItemInput
Output
DC BiasDC Bias Switch
Enclosure Material
FinishWeight
Size
OutlineSpecification
WR-15 Waveguide with UG-385/U Anti-Cocking Flange
WR-15 Waveguide with UG-385/U Anti-Cocking Flange
2.5 mm DC Jack (AC-to-DC power converter included)
On-Off Rocker Switch with Indicator Light
Extruded Aluminum
Black Anodized
1.5 lbs
2.36” (W) x 2.36” (L) x 4.10” (H)
TB-SV-A-C

www.sagemillimeter.com | 3043 Kashiwa Street, Torrance, CA 90505
Phone: 424-757-0168 | Fax: 424-757-0188 | Email: sales@sagemillimeter.com
Copyright © 2020 by SAGE Millimeter, Inc.

Page | 1

STB-5037533350-1515-L1-C

Advanced

50

5040403030Gain

2020Input Return Loss
Output Return Loss

101000-10-10-20-2050 51 52 53 54 55 56 57 58 59 60 61 62 63 64 65 66 67 68 69 70 71 72 73 74 75

Frequency (GHz)

Return Loss (dB)Gain (dB)

Typical Gain and Return Loss vs. Frequency

SAGE Millimeter, Inc.

Compact Bench Top Low Noise Amplifier, 50 to 75 GHz, 33 dB Gain, 5 dB NF

Typical Noise Figure vs. Frequency
65Noise Figure (dB)432

1050 51 52 53 54 55 56 57 58 59 60 61 62 63 64 65 66 67 68 69 70 71 72 73 74 75

Frequency (GHz)www.sagemillimeter.com | 3043 Kashiwa Street, Torrance, CA 90505
Phone: 424-757-0168 | Fax: 424-757-0188 | Email: sales@sagemillimeter.com
Copyright © 2020 by SAGE Millimeter, Inc.

Page | 2

STB-5037533350-1515-L1-C

Advanced

Mechanical Outline: (Unless otherwise specified, all dimensions are in inches [millimeters])

Note:
••
•
•
•

SAGE Millimeter, Inc.

Compact Bench Top Low Noise Amplifier, 50 to 75 GHz, 33 dB Gain, 5 dB NF

P1dB specification met by design and not tested.
All data presented is collected from a sample lot. Actual data may vary unit to unit.
All testing was performed under +25 ˚C case temperature.
Other mechanical configurations are available under different model numbers.
SAGE Millimeter, Inc. reserves the right to change the information presented without notice.

Caution:••••Exceeding absolute maximum ratings shown will damage the device.
The device is static sensitive. Always follow ESD rules when working with the device.
Any foreign objects in the waveguide will cause performance degradation and possible device
damage.
The case temperature of the device shall never exceed +50 ˚C. Use proper heatsink or fan if
necessary.

www.sagemillimeter.com | 3043 Kashiwa Street, Torrance, CA 90505
Phone: 424-757-0168 | Fax: 424-757-0188 | Email: sales@sagemillimeter.com
Copyright © 2020 by SAGE Millimeter, Inc.

Page | 3