Bone Grafting Product94

Sinus Lift Instrument

96

Periotome / Elevator

97

Measuring Device

98

Titanium Instrument

98

Resin Color Probe

99

93

Graft Carrier

Bone Scraper Type-2No. 30495(3 ea.)*Used for transfering of a autologous bone or an artificial bone under implant treatment.

*Collect own born like shaving allows less surgical invasive.
*Clear body allows how much you scraped own born inside to be visible.
*Suitable angle of the tip part allows easy to scrape bone.

Size / 155mm*Capacity / approx. 1cc*Width of blade / 6.4mm*Material / TPX

Bone Mill

Description

Diameter(mm)Capacity(cc)

30462

ø2.9

ø2.90.02

30463

ø3.4

ø3.40.03

Dissector-ex-2-*Used to mark on the gum before cut.

Name of Parts
No. 30480

No.

Pusher

No. 30020

Drum fixing pin

Grinding drum shaft

Shaft locking
screw

Base

1.4mm

Slot
CylinderGrinding drum
Stainless Cup
Ratchet handle*Able to comminute 0.1 to 1.0mm.
*Ratchet structure offers to comminute bones in justlight power.
*Simple structure makes shorten your time for thepreparation and cleaning of the operation.

*Size / 26 x 17mm (Slot)*Material / Radel R (Base)
Bakelite (Black part of handle)

94

95

Sinus Lift Instrument

Periotome*Able to operate extraction with minimum damage to surrounding periodontal tissues.

*Easy control angle design.

*Excellent for immediate implantation.

-Point-7.0mm

5.6mm

2.7mm

*Suitable for searching bone deeper inside maxillary sinus.

1.7mm*Smooth edges to prevent Schneider's membrane from damages.

#1#2

Anterior#3

Posterior

7m
1.

Anteriorm-Handle-4.0mm#A1

No. 30021
3.2mm

P-A (ø8mm)
4.0mm3.2mm

No.

Description

30016

Point #1

30017

Point #2

30018

Point #3

10201

P-A Handle#A2

No. 30022

Elevator-B Handle Spear-#A3*Since this product is an
instrument which cuts ap e r i o d o nt al li g am e nt an dc ar r ies out ex tr usion of atooth by wedge action, theblade is very thin. Therefore,
please do not perform
operation in which a root israised by leverage, there arepossibility of the breakage of
blade and fear of distortion.

* With a thin spear- like tip c an inser t in narrowp e r i o d o n t a l m e m b r a n e c a v i t y e a s i l y, a n dperiodontal ligament fibers are cut and it exfoliates.

No. 30023
4.0mm*Excellent for immediate implantation.

3.7mm

3.2mm

4.0mm

#B1

4.4mm

○No. 30031

#1S

8.0mm

10.0mmNo. 16141#2No. 16145

#B2

#B3

No. 30032#1C

No. 30033

×No. 16144

#3

#1R

No. 16146

No. 16147#B4

No. 30034[ #1C (Curve) ]4.4mm

No. 30035
10.0mm
10.0mm96[ #2 #3 ]4.4mm#B5[ #1R (Reverse) ]*The curve of the tip can approach the“molar” smoothly.

*Inside curve of the tip can approach the end ofmolar, lingual and palate side.

*This curved shape of a shank is suitably effective
for “molar” to approach with ease.
* The tip is positioned on the ex tended line of
the handle for handling balance. It makes Dr,scomfortable to keep the balance of operation.

97

Periodontal Color Probe Tip (Replaceable Type)

Parallel Meter-Tip (5 ea.)*A probe made of the resin prevents implant and a prosthesis being damaged.
*Black and yellow color combination offers easy distinction of the scale.
*Because of the external diameter is only φ0.45mm, this product can reduce the burdens to periodontium.

No. 21122#2No. 23522#4#5

No. 23524

No. 23525

a scaleMain body

X-Ray Gauge
Contents

a scale*Material / PPSU

Sleeve (2 ea.) / Gauge (2 ea.)*It's difficult to come off and remain long,
because of the laser marking permeates
inside the resin body.

-Handle-No. 21157

SleeveOctagonal
No. 13121

Gauge(1mm by 1mm with hold line every 5mm)

SC

*Size / W38 x D48 x H8.5mm (Sleeve)
W33 x D45mm (Gauge)*Material / Si (Sleeve)*Suitable for root canal treatment, accurate examination before
placing an implant, checking of calculus quantity and its area.

- Howto use - ②①

No. 13127③①スリーブの中に、②X線ゲージ③X線フィルムを装着して使用します。

YS

No. 13128

Periodontal Probe*Titanium

Green

No.13118

Blue

No.13119#5 Titanium
No. 09575

Pink*Material / POM

98

No.13120

99

Mouth Prop
*Able to keep the mouth open stably even for surgery which takes some hours especially under sedation.

-with Tongue GuardContents

Mouth Prop / Tongue Guard Right, Left
[ Without Tongue Guard ]LS

No.14676

No.14677
DHW

SS

No.14678

D*Material / Si (Main body)
PP (Tongue Guard)

Almighty Mouth Gag*Able to keep the mouth open stably even for surgery which takes some hours especially under sedation.

Main body, Tip (4 ea.)

Adult

No.14666*Bite pressure / up to 1400N*Material / Si (Tip)100

Size (mm)Description

W

Contents[ With Tongue Guard ]Mouth PropTongue GuardW

DHL

403517

S3529

16SS302516

L/S

52

34-

SS

39

24-