Scaler

14

Periodontal Curette

22

Explorer

29

Root Planing Check Scaler 29
Sharpening Product

32

Periodontal Probe

36

Resin Color Probe

38

Medicine Applicator

40

13Scaler-est2 Hygienist-*Efficiently remove calculus in all area with the combination of H5/33 and H6/H7.

Round

Sakura#H5/33#H5/33

Anterior(Interproximal area / Cervical area)

Anterior(Interproximal area / Cervical area)

No. 33308

No. 33300

Color Fit

#H6/H7#H6/H7

Anterior / Posterior(Mesial surface / Distal surface)*Handle diameter / ø9.5mm*Material / Si (ID Ring)*Handle diameter / ø9.5mm

No. 33309

No. 33301*Silicon handle offers softer grip.

Octagonal#H5/33#H5/33

Anterior(Interproximal area / Cervical area)

Anterior(Interproximal area / Cervical area)

No. 33310

No. 33302#H6/H7*Handle diameter / ø10.0 - ø12.0mm*Material / Si (Handle)

TOIRO

Anterior / Posterior(Mesial surface / Distal surface)

Anterior / Posterior(Mesial surface / Distal surface)

No. 33303#H6/H7*Handle diameter / ø6.5mm

Anterior / Posterior(Mesial surface / Distal surface)

No. 33311*Silicon handle offers softer grip.

#H5/33

Anterior(Interproximal area / Cervical area)

No. 33304#H6/H7*Handle diameter / ø9.0mm*Material / Si (Handle)

14

Anterior / Posterior(Mesial surface / Distal surface)

No. 33305

15

Scaler-est2 #204SD--est2 Slim*Combination of sickle and spoon tip.
*Suitable for scaling both neck and margin part of anterior.

Sakura*Handle diameter / ø9.5mm*Material / Si (ID Ring)

No. 33320*Handle diameter / ø9.5mm#Anterior
No. 33330

Color Fit

*Silicon handle offers softer grip.

*Handle diameter / ø10.0 - ø12.0mm*Material / Si (Handle)

TOIRO*Handle diameter / ø9.0mm*Material / Si (Handle)

No. 33321*Silicon handle offers softer grip.

No. 33323-Single EndRound#1

No. 09001*Handle diameter / ø9.5mm

No. 33322#3

No. 09003

Octagonal*Handle diameter / ø6.5mm#B

No. 09035#11DA

No. 09040#11DB

No. 09041

No. 33324#11SB*Specially designed crooked neck allow easy attempt against distal endof posterior teeth.

No. 09043#11 Slim

No. 09044#J1

No. 09111*The long neck also helps to reach distal end of last molar.

16

17

Scaler-Slim--CL-*Width of cutting blade is thinner than standard sickle scaler.

#1#1

No. 09151

Finishing for anterior(Crowded teeth / Narrow interproximal area)

No. 09411#2#3

No. 09152

No. 09153#2

Finishing for posterior(Mesial surface / Distal surface / Narrow interproximal area)-YB-

No. 09412#1*Specially designed crooked neckand long handle allow easy attemptagainst distal end of last molar.

*Since a cutting part is thinner thanregular sickle scalar, it is effective inthe place which is hard to operate.

Slim #1

No. 09101

Slim #2

Regular#2

No. 09102-Partner-#3

No. 09103#1

No. 09301-Ergonomics MK#2

No. 09302#3

No. 09303#2

No. 09352#3

No. 09353

18

19

Scaler

Scaler Tip (Replaceable Type)-for Furcation-*Remove all joint parts and replaceable tip before sterilization and disinfection.

*Specially designed for scaling and root planing at the furcation and deep subgingival area.

*For more replaceable type products see P. 211.

-Handle#F1

Anterior / Posterior

No. 09471

P-A (ø8mm)

P-B (ø8mm)

P-F (ø6mm)

P-C (ø6mm)

No. 10202

No. 10201

#F2

Lower

No. 09472

No. 10205

Diamond Coated File Scalers Nabers

No. 10203-Tip (3 ea.)-*Diamond coated on 360°.
*This feature makes it available to do operation withany part of the tip.
*Also can be effectively used for superior dome of thefurcation part and the corner part.

#11SB*Slide the tip part like "File" for an operation.

No. 23043

No. 09470*Material / Si (Handle)

Kawasaki #2
No. 23622

Scaler Tip (Replaceable Type)#11DAS

No. 23068#H5

No. 23072#11DBS

No. 23069#H33

No. 23073#2S for Lingual
No. 23053#H6

No. 23074*Remove all joint parts and replaceable tip before sterilization and disinfection.
*For more replaceable type products see P. 211.

#H7

-Handle-No. 23075

P-A (ø8mm)

TMDU #1(Tokyo Medical Dental University)

No. 23601

TMDU #3A(Tokyo Medical Dental University)

No. 23603

TMDU #3B(Tokyo Medical Dental University)

No. 23604

P-B (ø8mm)

No. 10201

No. 10202#J1

P-F (ø6mm)

No. 23036

P-C (ø6mm)

No. 10205#J4A

No. 23058#J5A

No. 23059#J4B

No. 23060

No. 10203-Tip (3 ea.)-

Slim #1A

No. 23076

20#B

No. 23035#11DA

No. 23040#11DB

No. 23041#J5B

No. 23061

21

SakuraAble to customize your curettewith ID Rings.
Variety of ID Ring is optional.

Gracey Curette and Scaler

series“est blade” is excellently sharpened with more time and effort than conventional ones.
Also water-sanding-process is applied so the hardness doesn’t become low through polishing.
We even designed the blade length slightly shorter to make it easier to control Gracey Curette and prevent thedamage with periodontal tissue thanks to the efficient operation.

Color FitThis soft and easygrip handle may
reduce your stressduring an operation.

Point

1

Sharp and Smooth blade
High-hardness material enables this “est”
to be long life blade

Point
Conventional one

est

Blade surface is not smooth enough due

Excellently smooth and flat surfaces maketo conventional sharpening process.

absolutely sharpened blade.

So it’s easier to be dull than “est” series.

This sharp blade and High-hardness material

Point"est"blade revealed!

2

Improved controllability thanks to slightly

make your hand comfortable

Round

This shorthandle is foreasy control andable to do UV
sterilization.

one

est

It was hard to control the blade inside theIt becomes much easier to control just by

narrow pocket under low visibility,

designing the blade slightly shorter.

It requires high technique to put and fit theblade on root surface.

4

Delicately designed handle toshort blade comparing with conventional

Conventionalone

22

you are looking forFind yourfavorite handle!

combination make the blade long life.

Point

3

Easy and fun to find what No.

Octagonal

Classic designof handle

TOIRO

This thin handle is suitablefor delicate operation.

23

Gracey CuretteSoft(Standard)*Neck flexibility : High*Best for finishing work of root
planing and scaling.

0. 8

5m

#G11-12m

0. 6 mPosterior (Distal surface)#G15-16#G13-14#G17-18m*Deep angled shank of #G15-16 and #G17-18prevent to be hit by patient's opposite teeth.

0.4mm

Hard

Hard

Posterior (Mesial surface)

0.55mm*Neck flexibility : Low*Best for scaling in deep narrow pockets,
furcation and crowding.

Mini

Short

Regular

Short*Neck flexibility : Low*Best for scaling of heavy and / orplenty of dental calculus.

Regular

Hard(Rigid)

Soft

Mini

Soft

Mini

#G1-2

#G3-4

#G5-6

#G7-8

#G9-10#G11-12#G13-14#G15-16#G17-18

#G1-2

#G3-4

#G5-6

#G7-8

#G11-12#G13-14#G15-16

Anterior

Anterior

Anterior
Premolar

Posterior

Posterior

Posterior

Posterior

Posterior

Posterior

Anterior

Anterior

Anterior
Premolar

Posterior

Posterior

Posterior

Posterior(Lingual surface
Buccal surface)(Furcation area ofLingual surface andBuccal surface )(Mesial surface)(Distal surface)(Mesial surface)(Distal surface)

Sakura(Lingual surface
Buccal surface)(Mesial surface)(Distal surface)(Mesial surface)

TOIRO

Hard

Description

Soft*Handle diameter / ø9.5mm*Material / Si (ID Ring)

#G1-2

Mini

Description

ID Ring

#G1-2

#G3-4

#G5-6

#G7-8

#G9-10#G11-12#G13-14#G15-16#G17-18

Hard

Green

33011

33012

33013

33014

33015

33016

33017

33018

33019

Soft

Pink

33021

33022

33023

33024

33025

33026

33027

33028

33029

Mini

Blue

33031

33032

33033

33034

-

33036

33037

33038

-

#G5-6

#G7-8

#G11-12#G13-14*Handle diameter / ø9.0mm*Material / Si (Handle)

#G1-2

#G5-6

HandleVioletYellow

Hard

33101

33103

#G7-8

#G11-12#G13-14

Blue

Pink

Green

33104

33106

33107

Soft

33111

33113

33114

33116

33117

Mini

33121

33123

33124

33126

33127

Description

#G1-2

#G5-6

#G7-8

#G11-12#G13-14

Hard

33071

33073

33074

33076

33077

Round

Color Fit
*Handle diameter / ø9.5mm

Soft

33081

33083

33084

33086

33087

Mini

33091

33093

33094

33096

33097

Octagonal
#G1-2

#G3-4

*Handle diameter / ø10.0 - ø12.0mm*Material / Si (Handle)

24#G9-10#G11-12#G13-14#G15-16#G17-18

#G5-6

#G7-8

Description

#G1-2

#G3-4

#G5-6

#G7-8

#G9-10#G11-12#G13-14#G15-16#G17-18

HandleVioletOrangeYellow

Blue

Red

Pink

Green

Navy

Lime

Hard

33041

33042

33043

33044

33045

33046

33047

33048

33049

Soft

33051

33052

33053

33054

33055

33056

33057

33058

33059

Soft

33141

33142

Mini

33061

33062

33063

33064

-

33066

33067

33068

-

Mini

33151

33152

*Handle diameter / ø6.5mm

Description

#G1-2

#G3-4

#G5-6

#G7-8

#G9-10#G11-12#G13-14#G15-16#G17-18

Hard

33131

33132

33133

33134

33135

33136

33137

33138

33139

33143

33144

33145

33146

33147

33148

33149

33153

33154

-

33156

33157

33158

-

25

Gracey Curette Tip (Replaceable Type)ID Ring*This ID Ring is only for Sakura handle.

*Remove all joint parts and replaceable tip before sterilization and disinfection.
*For more replaceable type products see P. 211.

-HandleViolet
Red*Diameter / ø7.4～9.8mm*Material / Si

OrangePink

Yellow
GreenNo.

Description

13690

Assort18pcs (9colors x 2)

13691

Violet

10pcs

13692

Orange

10pcs

13693

Yellow

10pcs

13694

Blue

10pcs

13695

Red

10pcs

13696

Pink

10pcs

13697

Green

10pcs

13698

Navy

10pcs

13699

Lime

10pcs

Blue
Navy

Lime

Q'ty

P-B (ø8mm)

P-F (ø6mm)

P-C (ø6mm)

No. 10202

No. 10201

No. 10205

No. 10203-Tip (3 ea.)*Select the color as you like to make it easy to findwhat you needed.

#G1

No. 23301#G5

No. 23305#G9

No. 23309

#G13

#G2

No. 23302#G6

No. 23306#G10

No. 23310#G3

No. 23303#G7

No. 23307

#G11

#G4

No. 23304#G8

No. 23308#G12

No. 23311

No. 23312#G14

No. 23313

No. 23314

Mini #G1

Mini #G2

Mini #G3

Mini #G4

Mini #G5

Mini #G6

Mini #G7

Mini #G8

Mini #G11

Mini #G12

Mini #G13

Mini #G14

No. 23321

No. 23325

No. 23331

26

P-A (ø8mm)

No. 23322

No. 23326

No. 23332

No. 23323

No. 23327

No. 23333

No. 23324

No. 23328

No. 23334

27

Periodontal CuretteRoot Planing Check Scaler*Double-end instrument with explorer and sickle scaler.

-KK-Type2-*It is suitable for the check of the left dental tartar and regular check.

Standard #1
Anterior

No. 09311

SC

No. 09478

ExplorerStandard #2
Posterior

No. 09312*Thinness and elasticity at the tip are effective in the check of dental tartar, cavity and surplus cement.

-Single EndMini #3

Anterior (Narrow Pocket)

No. 09313#3A

No. 01031-Double EndMini #4

Posterior (Narrow Pocket)

No. 09314*Material / Si (Handle)#11-12

No. 09224*Two cutting edges and internal angle 70° : Only two kinds of curette (anterior /posterior) can be done SRP in all area.

Blade Width2)
1, #m(# # 4)1.0m m(# 3,
0.7m

SRP for Posterior teeth-Gracey Curette-*Explorer with the shank of"Gracey Curette #11-12".

Buccal
Mesial

A：SRP
B：Gingival curettage

DistalLingualMesialDistalLingaul/Baccal：＃11・12：＃13・14：＃7・8/＃9・10

3 or 4 kinds of gracy
curettes are needed.

Debridement Scaler-Periodontal Curette KK-TypeBuccal

Ａ

Ｂ*Two cutting edges can be done both scaling and gingival curettage at
the same time.

28

Mesial

MesialLingaul/Baccal
DistalDistal#11-12

Lingual：＃2’s Inner blade：＃2’s Outer blade

Posterior teeth can bedone by one “Periodontal
curette KK-Type”.

No. 09227*This "DBS" is also effective for scaling narrow space,such asadjacent surface of molar part, and removing cement.
*Handle is marked "DBS"

29

Debridement InstrumentRound#11-12

No. 09225#11-12 Titanium
No. 09226#11-12

No. 09229*Handle is marked "DB #11-12".

*Temperately round edge is designed not to damage implant or prosthesis.
*Useable for the fundus of superstructure of an implant.

*Material / Si (Color Ring)*Handle diameter / ø9.5mm#11-12 Titanium*Material / Si (Color Ring)*Handle diameter / ø9.5mm

No. 09230＝ Debridement Instrument*Handle is marked "DB #11-12 Ti".

* Width of the edge part / 0.4mm

Explorer

ＤＢＩ*Temperately round edge is designed not to damage implant or prosthesis.
*Useable for the fundus of superstructure of an implant.

*Handle is marked "DB #11-12 Ti".

* Width of the edge part / 0.4mm

ＤＢＩ*Handle is marked "DB #11-12".

Gracey Curette[ Clinical Movie ]【 ＤＢＩ is effective for】
●Soft biofilm●Thin calculus●Firm gums at themaintenancestage

Debridement
Biofilm●Thin and thedelicate shapemake it easy to
do operation
gently .
●This flat-surface
inside the edge
remove thebiofilm
efficiently.

Fit
&
ch
Cat

●Gracey Curettes
have sharp edge
and are good for
SRP but
over-treatment
can be happened.

●Explorers do nothave blade so it
is hard to remove
the biofilm well.
Biofilm

Biofilm

Blade
part

Tip
part

Width
1/2

Width
1/2

●Approaching
to the narrow
space maydamage theperiodontal
tissues.

Biofilm

Blade part

Width
1

*Width :Ratio based on Gracey Curette as 1.

* Smooth approach to thenarrow space.

* Flat-surface inside the partcatch the biofilm excellently
while lifting.

* Soft-sharp edge do notdamage the prosthesis bydelicate touch.

* Good for removing the small* Easy access to narrow space

stains.

between the teeth.

[#11-12 Titanium]*Safety round edge for subgingival.
*Secure biofilm removing by flat and smooth surface

of inside the edge.

*Thin and flexible edge like an Explorer.

30

31

Sharpening Stone

Sharpener Re Born

Point 1 Switch your stone to Re Born

Fine Grit

Coarse Grit

Moving grind stone is designed at the angle of 70°.

Point 2

No. 13060

Anytime & AnywhereereA (Wedge)

F (Flat)Size / W100 x D26 x H6mm

No. 13001

Since it is very compact and cordless, it can use underscaling and for the vacant time easily anywhere.

Size / W100 x D26 x H6mm

FinishingNo. 13006-Arkansas-

Point 3 Stable resultre*Since arkansas sharpening stones are natural stones, product will differ from picture.

By mechanizing a motion and angle of a grind stone, user
can concentrate on an instrument to grind stone and thestable result is guaranteed.

Contents

Sharpening Stone / Scaling Tester

Contents

Sharpening Stone / Scaling Tester

Midium Grit

Midium Grit

Characteristic of Grind Stone
Ceramic Grind Stone*Fine Grit - Midium Grit

*For routine maintenancesharpening.

Diamond Grind Stone*Coarse Grit
*Sharpening of dull
instruments.

Contents

Product spec

Main Body
Ceramic Grind Stone
Diamond Grind Stone
Sharpening Tester
Instrument Cap
Angle Assist for Gracey
AA Alkaline battery (2 ea.)

[Main Body]*Size / W107 x D57 x H54mm*Weight / 170g (without battery)*Material / Plastic*Spining disc at the bottom*2 AA alkaline batteries operate the unit.

[Grind Stone]*Ceramic Grind Stone / Diamond Grind Stone*Requires no lubrication.
*Grindstone can be used for 4 times ofdifferent part of its surface.

6A (Wedge)

Gracey Curette

ScalerSize / W76 x D50 x H10mm

No. 13032

Midium Grit

-How to UseScaler

6AS (Wedge)

Size / W100 x D50 x H10mm

No. 13030

Gracey CuretteInstrument Oil*It prevents instruments from getting rust if you apply small amount of oil to the
instruments before autoclave sterilization.
* It is used as lubricating oil to the joint part of forceps as well.

4 (Flat)Size / W100 x D25 x H6mm

*It is also used as sharpening oil for scalers, curettes,and any other instrumentsl.

No. 13033-Ceramic (Wedge)Contents
No. 13044

1) Place the middle of the back of the
instrument blade and face shouldbe parallel on the platform.

2) Slightly move the instrument to moving grind
stone and the lateral side of the blade will
be against the stone for sharpening.

Midium Grit

20mL3) To sharpen the toe of the blade, turn the Re Born itself.

Size / W80 x D40 x H8mm

-How to use an"Angle Assist for Gracey"-"Angle assist for Gracey"

Terminal shank

Here is the face whichis colored orange.

This "Angle Assist for Gracey" isa guide which keeps inside of theedgeof Gracey Currete parallel to
Platform.

Then inside of the edge ofGracey Currete becomesparallel to Platform.
1) Put an "Angle assist for Gracey"
to Re Born.

32

No. 13093

Sharpening Stone / Scaling Tester

2) Keep the terminal shank of Gracey Curetteeven with the line of "Angle Assist forGracey" which is colored orange.

*No lubrication needed.

Sharpening Tester
No. 53016*V and U shaped grooved help
sharpening of toe of scaler andcurette.
*Size / ø8 x 100mm*Material / PC : Heatproof temperature 115°C*Alcoholic disinfection is possible.

33

How to Sharpening
Gracey Curette

Sickle Scaler
New Sickle Scaler

New Gracey Curette

Face

Face

Face

Face

Lateral face

70°- 80°

SectionWhy sharpen?

Sharpening stones*Improve calculus removal.

Comparison Chart*Improve tactile sensitivity.
*Minimize patient discomfort, reduce fatigue and save time.

70°- 80°

Basis

SectionLateral face

(Section)(Section)*Ideally, instruments should be sharpened before each use.

Name

OriginLubricant

Grit

Arkansas stone

NaturalOil

Fine - Midium

Sharpening stone

Synthetic

Water / Oil

CoarseCeramic stone

Synthetic-

Fine - Midium*Instruments should be kept sharp and true to their original design.

1) Apply a little sharpening oil on stone.
Stabilize the instrument on the table.
T h e f ac e of t h e b l a d e s h o u l d b eparallel to the floor.

1) Apply a little sharpening oil on stone.
Stabilize the instrument on the table.
The face of the blade should be parallelto the floor.
Shank*Arkansas stone : Finishing.

Points*Sharpening stone : Sharpening of dull instruments or re-contouring.

Shank*Sharpen with appropriate force.
*Apply a little sharpening oil on stone.
*Stabilize the instrument on the table.
*Keep appropriate angle of blade and stone.
*The face of the blade should be parallel to the floor.
*Firmly hold sharpening stone and instrument by hand.

BeforeAfter

BeforeSharpen here

70°- 80°70°- 80°2) Move the stone up and down withshor t strokes. The internal angle
should be 70° to 80°, when the stone
is correctly placed against the blade.
A b l a c k m et a l l i c p a r t i c l e a n d o i lgather in the surface of the blade.

Sharpening Oil
Oil is used to decrease friction heat and allows smooth sharpening.
Apply a few drops of oil on the stone and wipe with soft paper, then start
the sharpening.

After

Sharpen here

70°- 80°70°- 80°2) Move the stone up and down withshor t strokes. The internal angle
should be 70° to 80°, when the stone
is correctly placed against the blade.

(A sludge will appear )To assess the instrument
Check your sharpening by two methods below
1. Use a scaling tester

2. Check the reflected light on the edge

If the instrument has the sharp edge, you feel "bite" the blade into
the tester.

Face to the cutting edge and check the reflected light. If light reflects and it
is visible in the shape of a line, the edge does not sharp. Sharp edge does
not reflect light.

Sludge3) When you recognize the sludge on theblade, move gradually to the tip of theblade.

3) When you recognize the sludge
on the blade, move graduallyto the tip of the blade.
Repeat same method on theoter side.

*The side is finished up in the lune.

OKNG(If the side is finished in straight line, the point
becomes fragile)4) Atempt at inner face to remove
the roughened edge a few times
in a rolling motion.

To round the toe of a Curette, placestone at 45° to the face.

45°

OK

Edge
*Rotate the stone around the toe.

NG4) Atempt at inner face to remove
the roughened edge a few times
in a rolling motion.

*By following a simple sharpening routine, your instruments will reward you with a long trouble-free working life.
*Do not use the instrument with thinner blade or transformed by repeat sharpening.
*This is one of several sharpening technique.

3435Periodontal Probe

WHO (Superelastic Alloy)#1

No. 09556

No. 09510

WHO

No. 09551#2

No. 09550

CPITN-C
No. 09552#4

No. 09554

Flat

No. 09520#5No. 09555

Furcation
No. 09202*It is effective in diagnosis
of a root bifurcation area.

#5 Titanium
No. 09575*For measuring width and depth during placementof implant fixtures.

Periodontal Probe Tip (Replaceable Type)*Remove all joint parts and replaceable tip before sterilization and disinfection.

#6*For more replaceable type products see P. 211.

No. 09557-Handle-#7

No. 09558

P-A (ø8mm)

P-B (ø8mm)

P-F (ø6mm)

P-C (ø6mm)

No. 10202

No. 10201

Round

No. 10205#4

No. 09504#5

No. 09505*Handle diameter / ø9.5mm

36

No. 10203-Tip (3 ea.)#6

No. 09506#1

No. 23510#5

No. 23515

37Periodontal Color Probe Tip (Replaceable Type)

Measuring Instrument for Interdental Brush-Tip (5 ea.)*A probe made of the resin prevents implant and a prosthesis being damaged.
*Black and yellow color combination offers easy distinction of the scale.
*Because of the external diameter is only φ0.45mm, this product can reduce the burdens to periodontium.

#2

No. 23522

No. 23351#4#5

No. 23524
a scaleNo. 23525
a scaleMain body*It's difficult to come off and remain long,
because of the laser marking permeates
inside the resin body.
*Material / PPSU

*Material / PPSU

-Handle-*Please use for checking the size of interdental brush by dentists or dental hygienists.

Octagonal[ Clinical Movie ]No. 13121

SC

Notice(Please refer to the notices as follows before use.)

No. 13127* Please do not let the patients use this instrument by themselves.
* Please do not push interdental papillae.

YS* Please do not use for any other purposes.

-How to Use1) Set the checker to YDM Mirror Handle.

No. 13128
Enlarged scale of Measuring Instrument for Interdental Brush2) Insert the checker into interdental and read the scale.

Green

No.13118
3) Select the interdental brush which is one size smaller
than the space of interdental..

Blue

No.13119*Material / POM

38

Pink

No.1312039

Gram Probe#2

No. 09204#5

CPITN-C

No. 09203

No. 09205*Loading from 20 to 25g, the pin hits on the edge of slit.

Medicine Applicator

Double End
No. 09174

Holder Set
No. 15010

Contents

Holder / Tip

Individual Parts

40

No.

Description

15011

Tip

15022

Holder