par SYNC®
PNC

pollutant activity relay synchronization

World's First PNC + 4-gas + NOX Analyzer!

Flexible iPEMS: Design The Best Solution For YOU

CertifiedPNC
Sensor!

The 3DATX parSYNC® FLEX-PNC iPEMS (integrated Portable Emissions Measurement System) provides gaseous
AND nanoparticle measurements with a proprietary cartridge system. The gaseous side utilizes a “hot-swap”
cartridge system for traditional 4-gas measurement (CO2, CO, HC, O2) plus NOX (NO and NO2). The particulatematter side utilizes NMi, METAS, and PTB certified particle number concentration (PNC) sensor based on theextended diffusion charging principle.

Exhaust Gas >>>SensorSAmbient Air >>>FPFilterPump

Soot-Free PNC SolutionBenefits• No soot build-up• No flammable operating liquid• Insensitive to vibrations• No need of compressed air

New! Further customize your parSYNC FLEX-PNC to allow for real-time acquisition of:®

Wireless OBD Data Logger: User-defined ECU Data for LD and HD

Real-time GPS and Ambient Meteorology Data (pressure, temperature, humidity)
Ports for Additional Measurements (ie - exhaust flow rate, after-treatment temperature)
The parSYNC® Series of devices are powered by a common software interface, either directly on the imbedded LCD
screen or from a linked laptop, which provides a familiar and adaptable platform to each unique transportation
challenge presented.
The software and hardware embedded in parSYNC® is completely customizable and extremely valuable for the3DATX user community. This broad spectrum and flexibility of analytical and reporting functions for fleet
managers, manufacturers, consultants and regulatory compliance specialists is particularly useful as national andinternational Governments and Authorities ramp up new emissions standards.
Also available: Robust, field-ready, PNC calibrator with adjustable particle concentration range from0 to 107 #/cm3. Fast and simple setup. Includes salt generator.
501 John James Audubon
Suite 200
Buffalo, NY 14228
info@3DATX.com
1.844.303.3289

Thor Park 8300
3600 Genk Belgium
info@3DATX.com+32 2 639 39 39

3DATX.COM

SCAN ME!

FEATURES AND BENEFITS•••••••••••••

Small Size/Lightweight = Easily transported to job site▪ Size: 40 cm x 55 cm x 24 cm▪ Weight:◦ 16.5 kg (w/dual Milwaukee M18 8.0 Ah batteries)◦ 14.3 kg (w/o batteries)
Battery Life: More than 1.5 hours typically (20°Cambient, warm-up using wall-power)
Operation via interactive LCD display or from linked
laptopBuilt-in WiFi Access-point to transmit data in real-time tolaptopFully Automated Software (customizable for specificrequirements)
Simple and Quick Calibration Process (BAR97 Hi/Lo, etc)
Internal Power Supply: Two (2) 18V 8.0Ah Standard
Lithium-Ion Battery Packs
Measurement Cartridges with Monitored and Stabilized
Temperatures
Hot-Swap Replaceable Gas-side Sensor Cartridges (4-gasand NOX) eliminates downtime in the field
Built-in Exhaust Sample Condensate Removal
Volatile Particle Reduction
Rugged and Weather ResistantEasy to Maintain and Operate

SENSOR SPECIFICATIONS

1

Individual Electro-Chemical Cells
O2
NO2
NO

Gases

Non-Dispersive Infrared (NDIR)
CO2HC
CORange

0-20%

0-15%

0-4000 ppm10-5000 ppm

0-300 ppm

0-100%

T0-90 Time (s)

< 3.5

< 3.5

< 3.5<5< 35<6Resolution

0.01%

0.01%

1 ppm1-2 ppm

0.1 ppm

0.01%

AccuracyAbs± 0.3%± 0.02%± 8 ppm± 15 ppm± 5 ppm± 0.1%

Lorem

Rel± 3%

± 3%

± 3%

± 2%

± 2%

2%

Repeatability

Abs± 0.1%± 0.02%± 6 ppm± 5 ppm± 5 ppm± 0.1%

Rel

± 2%

± 2%

± 2%

± 2%

± 2%

± 2%

Propane mode, extends to 30,000 ppm (lower accuracy)

PNC

Details
Range
Resolution
Accuracy
Counting Efficiency

Data
5,000 - 5,000,000 #/cm3
1,000 #/cm3
25,000 #/cm3 absolute or ± 25% relative
23nm±5% : 20-60%50nm±5% : 60-130%80nm±5% : 70-130%
240104