FCONCREATION OF NEW VALUE

Product Catalogue

About Us

FCON CO., LTD.

CREATION OF NEW VALUE≪CREATION OF NEW VALUE≫

We aim to become a company that is continuously welcomed by society by providing products and services
that based on unique technology and new value creation.
Our main business includes design and development, manufacture and sale of mass flow controllers and theirapplied products.
We also keep challenging new business area.

FCON CO., LTD.
President and CEO
Kuniaki YAMANAKA【What is a Mass Flow Controller】
A mass flow controller is a device used to control the flow rate by measuring the mass flow rate of fluid and gases.
It is possible to control the flow rate stably with high accuracy because it is not required to be corrected due to thechange of temperature or used pressure.
Flow control by volume flow such as flow meters and needle valves, are required to be corrected due to thechange of temperature or pressure when they require a certain degree of the accurate flow rate.
Mass flow controllers and mass flow meters have been used for the field of semiconductor and flat panel displaythat require the high accuracy of measurement and control. In addition, they have been used in the wide variety
of processes such as in the field of analysis, fuel cell, liquid crystal, organic EL, biotechnology, food,environmental measurement, combustion gas control, factory equipment, laboratory equipment and etc.
We aim to design and manufacture products that can be used in a wide range of fields. Therefore, FCON's mass
flow controllers / mass flow meters are designed improving performance and reliability through originaltechnology pursuing basic performance.

【Principle and Structure of Mass Flow Controller】
The mass flow controllers are constructed from flow sensor, bypass, valve and the control circuit as shown in the
figure below.
In general, the flow sensor used is called thermal flow sensor. Two resistors generate heat when electric current
flows through these that are wounded on the upstream side and the downstream side, around the stainlesscapillary tube.
When no gas is flowing in the capillary tube, the heat of the upstream and downstream sides of the resistor is
kept equilibrium and the flow rate output signal shows zero.
Gas entering from the entrance is divided into the sensor and a bypass. When the gas begins to flow into thesensor, a temperature difference occurs between the upstream and the downstream resistors, thermal balance islost and the temperature distribution of the sensor changes. This change is captured by the bridge circuit as flowrate output signal. Compare the external flow rate setting signal (0 to 5V) and the flow rate output signal of thesensor, the flow control valve performs PID operation (proportional, integral, derivative) to match the signal level.
Since the valve opening is fine-tunes and adjust automatically, it is possible to control the flow rate under thesetting condition.

■Structure of Mass Flow Controller■Temperature distribution of Flow Sensor

CREATION OF NEW VALUE

Digital Mass Flow Controller
With Built-in Display

2000 series≪Features≫● Built-in display setting unit, setting control is possible only with the main unit.
● Driving power is DC24V single power supply (500mA or more AC adapter can be used).
● Compact size to realize small size and space saving.
● Simple installation to various equipment.
● Digital communication: RS485 communication.
● Analog signal: 0-5V and 4-20mA can be switched.
● Equipped with an abnormal flow rate alarm function.
● Equipped with accumulation function.
● Digital Control Application Software.
●Calibration certificates, calibration reports and traceability schemes
can be produced upon request for an additional fee.
【Specification】
Mass Flow Controller

C2005

C2020

Mass Flow Meter

M2005

M2020

Flow range (N2 equivalent)
Gases

10 SCCM～5SLM

Normally Closed (Excluding Mass Flow Meter)
2-100

Flow rate control range
Setting signal

20 SLM

Air, nitrogen, carbon dioxide, argon, hydrogen, helium, oxygen

Valve type
Accuracy

10 SLM±1

F.S. (Excluding Mass Flow Meter)

F.S. (Accuracy guaranteed between 15-35℃)

Setting section of main unit, Digital: RS485, Analog: 0-5V and 4.3-20mA can be switched(Excluding Mass Flow Meter)

Output signal

Main unit display: 7-segment LED, Digital: RS485, Analog: 0-5V and 4-20mA can be switched

Repeatability±0.2％ F.S.

Response speed
Operating differential pressure≦2sec
50-300 kPa

100-300 kPa

200-300kPa

Pressure resistance

1 MPa(G)Operating temperature

5～50℃（Accuracy guaranteed between 15～35℃）、≦85％RH（No condensation permitted）

Leak integrity

1×10-7Pa･m3/sec He

Mounting attitude

Not Specified (Free)

Drive power source+24VDC： ≧500mA

Communication

RS485, Analog: 0-5V and 4-20mA can be switched

Wetted surface material

SUS316, PTFE, Fluoro-Rubber

Seal material

Fluoro-Rubber

Actuator system

Solenoid (Excluding Mass Flow Meter)

Surface finishing

Machined finish

Standard fitting

1/4inch.SWL (equivalent) ※ For other fitting, please contact us.

1≪Dimensions≫All dimensions are in inches with [mm] in brackets≪Ordering≫C 2005 – 4S2 – 1L – N2（e.g.）
①

②

③

④

⑤

① Type C：Controller M：Meter② Flow range 2005（10 SCCM ～ 5 SLM） 2020（10 SLM, 20SLM）③ Fittings (*1) 4S2：1/4in.SWL(equivalent) KQ2：One-touch fittings④ Full scale(*2)
e.g.：50 SCCM→50 5 SLM→5L 10 SLM→10L⑤ Type of gas (*3) N2, Air,Ar,O2,H2,He,CO2 etc.
*1. For other fitting, please contact us.
*2. At FCON, flow rates (SCCM, SLM) are converted to values at 0℃, 101.3kPa abs (1atm) for calibration.
Please specify separately if you wish to calibrate at 20 ℃or 25 ℃.
*3. Gas type is an example; please contact us for other gases.

≪Signal cable≫（sold separately）●AC adapter for C2000：+24 VDC（DC22.8 ～ 25.2 V，500 mA and above）（recommend）●C2000 RS485 Communication Conversion Unit with Digital Control Application Software●C2000 RJ11 – RJ11Communication Cable●Analog Communication Cable
Please feel free to contact us for more details.
The contents of this catalog are subject to change without notice.

2

CREATION OF NEW VALUE

Analog Mass Flow Controller

1000 Series≪Features≫● It is an analog model specializing the basic functions of the mass flow controller.
●Performance and reliability are improved with basic functions that pursue original technology.
● Two types of Power Supply Control, Floor-standing "PA01S" and Panel Mount "PA01PS".
● Compact size.
● Suppression of Flow surge (overshoot) at the start (Requires more than 30 seconds of interval).
● Input & Output Signal：Analog (0～5VDC)● To control the mass flow controller, a separate control power supply and a signal cable are required.
●Calibration certificates, calibration reports and traceability schemes can be produced upon request for an
additional fee.
Option: Possible to manufacture a flow rate accuracy of ± 1% F.S.
【Specification】（※）under development

Mass Flow Controller (N.O.)

C1005

C1020

C1030

C1050

C1100

Mass Flow Controller (N.C.)

C1005 NC

C1020 NC（※）

C1030 NC（※）

C1050 NC（※）

-

M1005

M1020

M1030

M1050

-

10,20 SLM

30 SLM

50 SLM

100 SLM

Mass Flow Meter

10,20,30,50,100,200
Flow range (N2 equivalent)

300,500 SCCM
1,2,3,5 SLM

Gases
Valve type

N2, Air, Ar,O2,H2,He,CO2 etc.
NO: Normally Open

NC: Normally Closed (Excluding Mass Flow Meter)

5-100

Flow rate control range
Accuracy

Air・N2(Please consult with us about corrosive gas)

±2

F.S (Excluding Mass Flow Meter)

F.S (Accuracy guaranteed between 15-35℃)

Setting signal0.25-5VDC (Excluding Mass Flow Meter)

Output signal

0-5VDC

Repeatability
Response speed
Operating differential pressure

±0.2

F.S.

±0.5

≦6 sec≦10 sec50-300 kPa（C1020NC、100～300kPa）

F.S.
≦15 sec

200-300 kPa

Pressure Resistance

1 MPa(G)Operating temperature

5～50℃（Accuracy guaranteed between 15～35℃）、≦85％RH（No condensation permitted）
1×10-7Pa･m3/sec He

Leak Integrity
Mounting attitude

Not Specified (Free)

Drive power source+15VDC：60mA、-15VDC：150mA

Communication

Analog 0-5 VDC

Wetted surface material

SUS316，PTFE，PEEK，Fluoro-Rubber

Seal material

Fluoro-Rubber

Actuator system

Thermal (Excluding Mass Flow Meter)

Surface Finishing

Machined finish

Standard fittings

1/4in.SWL (equivalent), One-touch ﬁttings,

3※ For other fitting, please contact us.

≪Dimensions≫

C1005/M1005

C1020～1050/M1020～1050/C1100
All dimensions are in inches with [mm] in brackets.

≪Ordering≫

C 1005 NC – 4S2 – 5L – N2（e.g.）
①

②

③

④

⑤

⑥

① Type C:Controller M:Meter② Flow range 1005(10 SCCM ～ 5 SLM ) 1020(10 ～ 20 SLM) 1030(30 SLM)
1050(50 SLM) 1100(100 SLM)
③ Valve type w/o :Normally open NC :Normally closed (1100 not covered)④ Fitting (*1) 4S2:1/4in.SWL(equivalent) 6S2:3/8in.SWL(equivalent) KQ2:One-touch fittings⑤ Full scale (*2) e.g. :50 SCCM→50 5 SLM→5L⑥ Gas types (*3) N2, Air,Ar,O2,H2,He,CO2 etc.
*1. For other fitting, please contact us.
*2. At FCON, flow rates (SCCM, SLM) are converted to values at 0℃, 101.3kPa abs (1atm) for calibration.
Please specify separately if you wish to calibrate at 20 ℃or 25 ℃.
*3. Gas type is an example; please contact us for other gases.

≪Signal cable≫
When purchasing the mass flow controller, it is possible for us to produce signal cables and replacement cables for the
power supply control currently in use. In the situation whereby the shape of the signal cable connector from theequipment is different, if the information of the connector shape and pin assignment is available, we can manufacture
the signal cables in addition to the standard products. Depending on the type of connector, it may be difficult tomanufacture. Please feel free to contact us for more details.
The contents of this catalog are subject to change without notice.

4

CREATION OF NEW VALUE

Control Unit

PA01PS
PA01S≪Features≫●Cost effective and easy to use.
●Floor-standing "PA01S" and Panel Mount "PA01PS".
●Compact size.
●Flow rate display and setting functions.
●Mass flow power supply：±15 VDC●Input/output：Analog Signal (0 ～ 5 VDC)
●Output voltage (0 ～ 5 VDC) output terminal is at the back of the unit as standard set up.
【Specification】
Model

PA01PS（Panel mount type）/ PA01S（Floor standing type）

Input Power

AC100 - 240 V

Mass flow power

0～5 VDC：Setting Signal +15 VDC：300 mA -15 VDC：200 mA

Mass flow control

10-turn potentiometer

Mass flow output

0～5 VDC

Flow rate setting / output display

0～100％ (*1)

Display switching (setting/output)

Toggle switch

Connector

D-sub9Pin female

Operating environment

5～50℃（Accuracy guaranteed between 15～35℃）、≦85％RH（No condensation permitted）

Mass flow input/output

0～5 VDC（Linked to the display switching toggle switch）*1. Flow rate display setting changes is possible.

≪1000 series connection≫
Cables（Optional）
Cables（Optional）

5

≪Dimensions≫
PA01PS

PA01S

All dimensions are in inches with [mm] in brackets.

≪Analog Cables SC0x（Optional）≫



Signal Cable：MFC side D-sub9 female connector ⇔ Power supply side D-sub9 male connector
You can order the standard 1 meter, 2 meters, and 3 meters length cable.
Custom-length cable is available.

The contents of this catalog are subject to change without notice.

6

CREATION OF NEW VALUE

Mass Flow Control / Measurement Unit

CUBE MFC≪Features≫● Compact and lightweight integral unit.
● Easy flow measuring and controlling● Flow rate display: Display (SET / OUT selector switch)●Output voltage (0 - 5 VDC) output terminal is at the back of the unit as standard set up.
●Upon request, Calibration certificate, Calibration report, and Traceability scheme can be prepared withadditional charges.
Option: Possible to manufacture a flow rate accuracy of ± 1% F.S.

【Specification】
Model ：Controller（N.O）
Model ：Controller（N.C）

CUBE MFC 1005／1020／1030／1050／1100
CUBE MFC 1005NC （※ under development : 1020NC／1030NC／1050NC）

Model ：Meter（M）

CUBE MFM 1005／1020／1030／1050

Mass Flow Controller (*1)

1000 series

Flow range
Operating temperatureStandard fitting

10 SCCM～100 SLM
5～50℃（Accuracy guaranteed between 15～35℃）、≦85％RH（No condensation permitted）
1/4in.SWL (equivalent), One-touch ﬁttings,

Setting signal/Mass flow control※ For other fitting, please contact us.

0.25～5 VDC 10-turn potentiometer

Output signal

0～5 VDC(*2)

Flow rate setting / output display

0～100

Display switching (setting / output)

Toggle Switch

Output terminal

Setting signal/Output signal : 0～5 VDC

Input power

AC100-240 V*1. Mass flow controller is conform to 1000 series specification.
*2. Flow rate display setting changes is possible.

Can be used both horizontally and vertically7≪Dimensions≫All dimensions are in inches with [mm] in brackets≪Ordering≫

CUBE MF C 1005 NC – 4S2 – 1L – N2 (e.g.)
①

②

③

④

⑤

⑥

① Type C:Controller M:Meter② Flow range 1005(10 SCCM～5 SLM) 1020(10～20 SLM) 1030(30 SLM)
1050(50 SLM) 1100(100 SLM)
③ Valve type w/o: Normally open NC: Normally closed (1100 not covered)④ Fitting4S2 1/4inch.SWL(equivalent) KQ2 One-touch fittings⑤ Full scale (*1) e.g. :50→50 SCCM 5L→5 SLM⑥ Gas types (*2) N2, Air,Ar,O2,H2,He,CO2 etc.
*1. At FCON, flow rates (SCCM, SLM) are converted to values at 0℃, 101.3kPa abs (1atm) for calibration.
Please specify separately if you wish to calibrate at 20 ℃ or 25 ℃.
*2. Gas type is an example; please contact us for other gases.

8

CREATION OF NEW VALUE
The contents of this catalog are subject to change without notice.

Gas flow Meter

GFM Series≪Features≫● Equipped with a mass flow sensor● Supports up to 200 LM in compact size
● Low pressure loss
● Flow output signal：Analog 0 - 5VDC
● A dedicated power supply and signal cable are required【Specification】
Model
Flow range

GFM050

GFM100

GFM20050 LM

100 LM

200 LM

Gases

Air , N2

Flow rate range

0 - 100% F.S.

Flow rate accuracy

0 - 19% ± 3% F.S. , 20 - 100% ± 2% F.S. (25℃)

Repeatability± 0.5% F.S.

Response speed≦10 sec

Pressure resistance

1 MPa(G)Flow output signal

Analog: 0 - 5 VDC

Operating temperature

5 - 50℃（Accuracy guaranteed between 15 - 35℃）、≦85% RH（No condensation permitted）

Mounting attitude

Not Specified (Free)

Drive power source± 15VDC ± 5% (DC 40mA)

Communication

Analog 0 - 5 VDC

Seal material

Fluoro-Rubber

Wetted surface material

Aluminum(A6061),SUS316,SS(Trivalent chromate plating),Fluoro-Rubber

Standard fittings

3/8R（Male） SS(Trivalent chromate plating)

Weightapprox. 300g

≪Example of wiring≫
Optional：A dedicated power supply and signal cable are required
Signal cable：［SC – M – ＊］［SC – DPM – ＊］AC100 - 240V

Digital panel meter

9Power supply：PA02M

≪Dimensions≫All dimensions are in inches with [mm] in brackets.

≪Ordering≫

GFM050 – 3/8R – 50ＬＭ – N2 - 4
①

②

③

④

⑤

① Model /Flow range GFM050(50LM) GFM100(100LM) GFM200(200LM)② Fitting 3 / 8R（male）③ Full scale 50 LM 100 LM 200 LM④ Gas types Air Nitrogen (N2)⑤ Connection pipe diameter(*1) 4：1/4 inch. 6：3/8 inch. 8：1/2 inch. 4M：4ｍｍ 6M：6ｍｍ 10M：10ｍｍ*1

Please specify the pipe diameter of the gas line to be installed.

※

AT FCON, flow rates(LM,CCM) are converted to values at25℃, 101.3kPa abs (1atm) for calibration.

The contents of this catalog are subject to change without notice.

10

CREATION OF NEW VALUE

PA02M Power supply(GFM dedicated power supply)≪Features≫● Compact size
（W135 ×H60 (Excluding protrusions) ×D60mm）● 5 line connection※Please use within the power output range.
● Meter (GFM) power supply: ± 15 VDC● Display power supply ：+ 5 VDC【Specification】
Model
GFM power
supplyOutputDisplaypower
power
supplyNumber of connection
Flow output signal
Connector
Operating environment
Input Power

PA02M＋15 VDC：0.3A －15 VDC：0.2A＋5 VDC：2A
Maximum：5
Analog 0 - 5 VDC※Output to the display when the cable is connected.
B5B-XH-A(LF)(SN)：Display (5 core)
B6B-XH-A(LF)(SN)：GFM (6 core)5 - 50℃ 、≦85%RH（No condensation permitted）
AC100 - 240 V≪Dimensions≫All dimensions are in inches with [mm] in brackets.
※Fixing screw (M3): The screwing depth into the housing should be up to 2 mm.

11≪Optional≫

Signal cable between GFM and PA02M
Model : SC - M - ＊Signal cable between PA02M and Digital panel meter
Model : SC - PM - ＊Digital panel meter with signal cable
Model : SC - DPM - ＊※ Standard length of signal cable 1 , 2 , 3 m (More than 4m is order made.)※ ＊The choice of cable length.
e.g.：SC - M - 1M

Model＊

Cable length

1M

1m

2M

2m

3M

3m

The contents of this catalog are subject to change without notice.

12

CREATION OF NEW VALUE

Economical Gas Mixer

CUBE GM series≪Features≫● All-in-one unit with mass flow controller, control power supply, valve, check valve and gas mixer● This gas mixer unit type consists of mass flow controller C1000 series and power supply control PA01PS.
● Equipped with a mass flow controller to supply a stable gas mixture that is not easily affected by pressure
and temperature.
【Specification】
Model

CUBE GM2（two gases）／CUBE GM3（Three gases）／CUBE GM4（Four gases）

Built-in MFC

C1000 series（manufactured by FCON）or FCST1000 series（manufactured by Fujikin）

Flow range (*1)≦50SLM（Total flow rate）

Flow rate control range

Conform to built-in mass flow controller specification

Operating differential pressure

Conform to built-in mass flow controller specification

Pressure Resistance

Conform to built-in mass flow controller specification

Operating environment

5～50℃（Accuracy guaranteed between 15～35℃），85 RH（No condensation permitted）

Wetted surface material

SUS316，PTFE，Fluoro-Rubber and built-in mass flow controller material

Standard Fitting

1/4inch.SWL (equivalent) ※For other fitting, please contact us.

Mass flow control

Conform to PA01PS power supply control(*2)

Flow rate setting/output display

0～100

Input power

AC100-240 V*1. At FCON, flow rates (SCCM, SLM) are converted to values at 0℃, 101.3kPa abs (1atm) for calibration.
*2. Flow rate display setting changes is possible.

【Flow Diagram（e.g.：3 gases）】

13

≪Dimensions≫

CUBE GM2：W315×D298×H250(mm)CUBE GM3：W415×D298×H250(mm)CUBE GM4：W515×D298×H250(mm)※Not including protrusions.
※Not including protrusions.
※Not including protrusions.

≪Ordering≫

CUBE GM3①① Mixing gases (*1)
GM2 Two gases
GM3 Three gases
GM4 Four gases*1. More than four gases, please contact us.
We can design and manufacture upon request in addition to standard products,

The contents of this catalog are subject to change without notice.

14

CREATION OF NEW VALUE

Touch panel Gas Mixer

CUBE GM-X series≪Features≫● All-in-one unit with mass flow controller, control power supply, valve, check valve and gas mixer●Program operation：Flow rate and time is possible to set for each line with program.
● Alarm function：Based on each line upper and lower limit setting, alarm sound if flow rate abnormality occurs● Unit type gas mixer● Equipped with a mass flow controller to supply a stable gas mixture that is not easily affected by pressure
and temperature.
【Specification】
Model
Built-in MFC

CUBE GM-X2（Two gases）／CUBE GM-X3（Three gases）／CUBE GM-X4（Four gases）
C1000series／Ｃ2000series(manufactured by FCON) or FCST1000series（manufactured by Fujikin）

Flow range (*1)

≦50 SLM（Total flow rate）

Flow rate control range

Conform to built-in mass flow controller specification

Operating differential pressure

Conform to built-in mass flow controller specification

Pressure Resistance

Conform to built-in mass flow controller specification

Operating environment

5～50℃（Accuracy guaranteed between 15～35℃），85％RH（No condensation permitted）

Wetted surface material

SUS316，PTFE，Fluoro-Rubber and built-in mass flow controller material

Standard Fitting

1/4inch.SWL (equivalent) ※For other fitting, please contact us.

Mass flow control

Touch panel operation

Input power

AC100-240 V*1. At FCON, flow rates (SCCM, SLM) are converted to values at 0℃, 101.3kPa abs (1atm) for calibration.

【Flow Diagram（e.g.：3 gases）】

15

≪Dimensions≫

CUBE GM-X2：W350×D400×H280(mm)CUBE GM-X3：W430×D400×H280(mm)CUBE GM-X4：W510×D400×H280(mm)※Not including protrusions.
※Not including protrusions.
※Not including protrusions.

≪Ordering≫

CUBE GM-X3①① Mixing gases (*1)
GM-X2 Two gases
GM-X3 Three gases
GM-X4 Four gases*1. More than four gases, please contact us.
We can design and manufacture upon request in addition to standard products,

The contents of this catalog are subject to change without notice.

16

CREATION OF NEW VALUE

Touch panel Gas Mixer（Buffer tank specification）

CUBE GM-Xα series≪Features≫● All-in-one unit with mass flow controller, control power supply, valve, check valve and gas mixer●Program operation：Flow rate and time is possible to set for each line with program.
● Alarm function：Based on each line upper and lower limit setting, alarm sound if flow rate abnormality occurs● The buffer tank pressure is constantly monitored, when the pressure falls below a certain level, gas is
automatically supplied to the buffer tank at the flow rate set (automatic stop and automatic restart pressure
are fixed values). Easy to move with equipped caster with brake.
● Equipped with a mass flow controller to supply a stable gas mixture that is not easily affected by pressure
and temperature.
【Specification】
Model

CUBE GM-Xα2（Two gases）／CUBE GM-Xα3（Three gases）／CUBE GM-Xα4（Four gases）

Built-in MFC

C1000series／C2000series(manufactured by FCON) or FCST1000series（manufactured by Fujikin）

Flow range (*1)

≦50 SLM（Total flow rate）

Flow rate control range

Conform to built-in mass flow controller specification

Operating differential pressure

Conform to built-in mass flow controller specification

Pressure Resistance

Conform to built-in mass flow controller specification

Operating environment

5～50℃ (Accuracy guaranteed between 15～35℃), 85％RH（No condensation permitted）

Wetted surface material

SUS316, PTFE, Fluoro-Rubber

Standard Fitting

1/4inch.SWL (equivalent) ※For other fitting, please contact us

Mass flow control

Touch panel operation

Tank volume (*2)

10 L, 20 L

Tank operating pressure

0.49 MPa 以下

Input power

AC100-240 V*1. At FCON, flow rates (SCCM, SLM) are converted to values at 0℃, 101.3kPa abs (1atm) for calibration.
*2. Other tank volume please kindly discuss with us.

17≪Dimensions≫

CUBE GM-Xα2：W350×D400×H1100(mm)CUBE GM-Xα3：W430×D400×H1100(mm)CUBE GM-Xα4：W510×D400×H1100(mm)※Not including protrusions.
※Not including protrusions.
※Not including protrusions.

≪Ordering≫

CUBE GM-Xα3①①

Mixing gases (*1)

GM-Xα2 Two gases
GM-Xα3 Three gases
GM-Xα4 Four gases*1. More than four gases, please contact us.
We can design and manufacture upon request in addition to standard products,

The contents of this catalog are subject to change without notice.

18

Company Profile
We are the manufacturer of flow control devices such as Mass Flow Controllers (MFC).
We have been providing wide range of flow control devices catering to our customers’ needs. Fromdesigning and manufacturing of standard and customize devices by our own.
Mass flow controllers are the devices to measure the mass flow of gases and liquids in order to regulate
the flow, indispensable for the processes requiring high-precision flow measurement. Because of the highprices of these devices, however, their uses have been confined to specific fields.
Our mass flow controllers are the devices that realizing the reasonable cost while retaining their basic
functions. They can be used for wide variety of fields ranging from general industry, mounts for various
kinds of production devices, and for research and development.
≪Management Philosophy≫ New value and market creation

We are creating a new market, which products and services from the user’s perspective by ideas and creation
which is not particular about the existing concept.
 Customer satisfaction is our existence value

We raise our existence value by providing valuable products and service for customers.
 Work environment with hopes and dreams

To create a rewarding workplace where employees are always motivated and fully demonstrate their potentials,
have dreams and hopes for the future.
≪Company Profile≫
Company Name

FCON, CO., Ltd.

President and CEO

Kuniaki Yamanaka

Establishment

July 2013

CapitalJPY 10,000,000

Business Content

Manufacturing and sale of various flow controllers

Head Office#109 Nankoku Office Park Center, 1-1-1 Hotarugaoka,
Nankoku-shi, Kochi, 783-0060, JapanTel＋81-88-855-7100

Fax＋81-88-855-7166

E-Mailcontact@fcon-inc.jp

URL

http://www.fcon-inc.jp/

【MEMO】

230405FCON
FCON CO., LTD.
1-1-1-109 Hotarugaoka,
Nankoku, Kochi, JAPAN
783-0060
TEL: ＋81-88-855-7100
http://www.fcon-inc.jp/