UHPLC & HPLC HILIC column
SPE

新世代ヒリックカラム

Website: www.hilicon.com

株式会社 クロマニックテクノロジーズ

3種類の固定相を提供

ヌクレオチドの分離

人工甘味料の分離

iHILIC®-Fusion
AMP

iHILIC®-Fusion(+)

ADPATP

NaphthaleneiHILIC®-Fusion(+)
AMP
ADP

Naphthalene

ATP

Column: 100 × 2.1 mm, 3.5-µm 100ÅMobile phase: A) 100 mM ammonium formate,
pH = 5.8 / acetonitrile = 30 /70
Flow rate: 0.2 mL/min
Column temperature: AmbientInjection volume: 2 µL

LC–MS–MS system: Advance UHPLC with a CTC
PAL autosampler and Bruker EVOQ™ Elite triple
quadrupole system with electrospray ionization;multiple reaction monitoring mode (MRM) in ESI(+)
and ESI(-) was used for selective and sensitive
detection, see details in Table.
Column: 150 × 2.1 mm, 3.5-µm 100ÅiHILIC-Fusion(+)
Gradient elution: A) acetonitrile; B) 10 mM
ammonium formate, pH = 3.5; gradient elution
from (95/5) A/B to (84/16) A/B in 8.5 min.
Flow rate: 0.3 mL/min
Column temperature: 40 °C
Injection volume: 5 µL

アミノ酸の分離標準品
iHILIC®-Fusion(+)

ダイエットサプリ

LC–MS System: Agilent 1100er LC system andThermo Fisher LTQ™ equipped with a HESI source,operated in positive ionization mode for analysis ofstandards. For the dietary supplement, anOrbitrap™ Exactive classic equipped with a HESI
source and operated in positive ionization mode.
Column: 150 × 2.1 mm, 3.5-μm iHILIC-Fusion(+)
Gradient Elution: A) acetonitrile–water–1 M
ammonium acetate, pH 5.75 (90:5:5); B) water–acetonitrile–1 M ammonium acetate, pH 5.75
(90:5:5); 0–0.5 min (90:10) A–B; 0.5 to 25 min,gradient elution from (90:10) A–B to (60:40) A–B.
Flow Rate: 0.3 mL/min
Column Temperature: 40 °C
Injection Volume: 5 μL
Amino Acids: Arginine, asparagine, aspartic acid,glutamic acid, glutamine, hydroxyl-proline,
isoleucine, leucine, lysine, phenylalanine, proline,
tryptophan, tyrosine, and valine. 50 μM of eachamino acid was dissolved in water–acetonitrile(25:75) solution.

リン脂質の分離リゾビスフォスファチジン酸

ホスファチジン酸

ホスファチジルグリセロール

iHILIC®-Fusion

カルジオリピンホスファチジルイノシトール

ホスファチジルコリン

ホスファチジルエタノールアミン

リゾホスファチジルコリン

ホスファチジルセリン

リゾホスファチジン酸

リン脂質の化学構造

LC–MS system: Thermo Scientifi c Ultimate 3000 HPLC
system coupled to a Q Exactive™ Plus HybridQuadrupole-Orbitrap™ mass spectrometer, which isequipped with a HESI II source and operated in negative
ionization mode (ESI-).
Column: 150 × 2.1 mm, 3.5 μm 100 Å iHILIC -Fusion.
Eluent:A) ammonium acetate solution (35 mM, pH 5.75) andacetonitrile (95:5, v/v);
B) acetonitrile.
Gradient elution: 0–0.5 min, 97% B; 0.5–26.5 min, from
97% to 75% B; 27–33 min, 60% B; 35–45 min 97% B.
Flow rate: 0.3 mL/min
Column temperature: 40 °C
Injection volume: 10 μL®

【 iHILIC-Fusion カラムおよびSPEカートリッジの価格表 】P/NiHILIC®-Fusion UHPLC Column, SS – Stainless steel column hardware

価格

P/NiHILIC®-Fusion(+) UHPLC Column, SS – Stainless steel column hardware

価格110.052.0110

iHILIC®-Fusion HILIC Column, SS, 50x2.1 mm, 1.8 µm, 100Å

¥120,000

100.052.0110

iHILIC®-Fusion(+) HILIC Column, SS, 50x2.1 mm, 1.8 µm, 100Å

¥120,000

110.102.0110

iHILIC®-Fusion HILIC Column, SS, 100x2.1 mm, 1.8 µm, 100Å

¥130,000

100.102.0110

iHILIC®-Fusion(+) HILIC Column, SS, 100x2.1 mm, 1.8 µm, 100Å

¥130,000

110.152.0110

iHILIC®-Fusion HILIC Column, SS, 150x2.1 mm, 1.8 µm, 100Å

¥143,000

100.152.0110

iHILIC®-Fusion(+) HILIC Column, SS, 150x2.1 mm, 1.8 µm, 100Å

¥143,000

110.053.0110

iHILIC®-Fusion HILIC Column, SS, 50x3.0 mm, 1.8 µm, 100Å

¥128,000

100.053.0110

iHILIC®-Fusion(+) HILIC Column, SS, 50x3.0 mm, 1.8 µm, 100Å

¥128,000

110.103.0110

iHILIC®-Fusion HILIC Column, SS, 100x3.0 mm, 1.8 µm, 100Å

¥135,000

100.103.0110

iHILIC®-Fusion(+) HILIC Column, SS, 100x3.0 mm, 1.8 µm, 100Å

¥135,000110.153.0110

iHILIC®-Fusion HILIC Column, SS, 150x3.0 mm, 1.8 µm, 100Å

¥151,000

100.153.0110

iHILIC®-Fusion(+) HILIC Column, SS, 150x3.0 mm, 1.8 µm, 100Å

¥151,000P/N

iHILIC®-Fusion HPLC Column, PEEK – Engineering plastic column hardware

価格

P/N

iHILIC®-Fusion(+) HPLC Column, PEEK – Engineering plastic column hardware

価格110.022.0310

iHILIC®-Fusion HILIC Column, PEEK, 20x2.1 mm, 3.5µm, 100Å

¥90,000

100.022.0310

iHILIC®-Fusion(+) HILIC Column, PEEK, 20x2.1 mm, 3.5µm, 100Å

¥90,000

110.052.0310

iHILIC®-Fusion HILIC Column, PEEK, 50x2.1 mm, 3.5µm, 100Å

¥102,000

100.052.0310

iHILIC®-Fusion(+) HILIC Column, PEEK, 50x2.1 mm, 3.5µm, 100Å

¥102,000

110.102.0310

iHILIC®-Fusion HILIC Column, PEEK, 100x2.1 mm, 3.5µm, 100Å

¥110,000

100.102.0310

iHILIC®-Fusion(+) HILIC Column, PEEK, 100x2.1 mm, 3.5µm, 100Å

¥110,000

110.152.0310

iHILIC®-Fusion HILIC Column, PEEK, 150x2.1 mm, 3.5µm, 100Å

¥128,000

100.152.0310

iHILIC®-Fusion(+) HILIC Column, PEEK, 150x2.1 mm, 3.5µm, 100Å

¥128,000

110.054.0310

iHILIC®-Fusion HILIC Column, PEEK, 50x4.6 mm, 3.5µm, 100Å

¥104,000

100.054.0310

iHILIC®-Fusion(+) HILIC Column, PEEK, 50x4.6 mm, 3.5µm, 100Å

¥104,000

110.104.0310

iHILIC®-Fusion HILIC Column, PEEK, 100x4.6 mm, 3.5µm, 100Å

¥123,000

100.104.0310

iHILIC®-Fusion(+) HILIC Column, PEEK, 100x4.6 mm, 3.5µm, 100Å

¥123,000

110.154.0310

iHILIC®-Fusion HILIC Column, PEEK, 150x4.6 mm, 3.5µm, 100Å

¥143,000

100.154.0310

iHILIC®-Fusion(+) HILIC Column, PEEK, 150x4.6 mm, 3.5µm, 100Å

¥143,000

110.022.0510

iHILIC®-Fusion HILIC Column, PEEK, 20x2.1 mm, 5µm, 100Å

¥89,000

100.022.0510

iHILIC®-Fusion(+) HILIC Column, PEEK, 20x2.1 mm, 5µm, 100Å

¥89,000

110.052.0510

iHILIC®-Fusion HILIC Column, PEEK, 50x2.1 mm, 5µm, 100Å

¥97,000

100.052.0510

iHILIC®-Fusion(+) HILIC Column, PEEK, 50x2.1 mm, 5µm, 100Å

¥97,000

110.102.0510

iHILIC®-Fusion HILIC Column, PEEK, 100x2.1 mm, 5µm, 100Å

¥104,000

100.102.0510

iHILIC®-Fusion(+) HILIC Column, PEEK, 100x2.1 mm, 5µm, 100Å

¥104,000

110.152.0510

iHILIC®-Fusion HILIC Column, PEEK, 150x2.1 mm, 5µm, 100Å

¥120,000

100.152.0510

iHILIC®-Fusion(+) HILIC Column, PEEK, 150x2.1 mm, 5µm, 100Å

¥120,000

110.252.0510

iHILIC®-Fusion HILIC Column, PEEK, 250x2.1 mm, 5µm, 100Å

¥147,000

100.252.0510

iHILIC®-Fusion(+) HILIC Column, PEEK, 250x2.1 mm, 5µm, 100Å

¥147,000

110.054.0510

iHILIC®-Fusion HILIC Column, PEEK, 50x4.6 mm, 5µm, 100Å

¥99,000

100.054.0510

iHILIC®-Fusion(+) HILIC Column, PEEK, 50x4.6 mm, 5µm, 100Å

¥99,000

110.104.0510

iHILIC®-Fusion HILIC Column, PEEK, 100x4.6 mm, 5µm, 100Å

¥117,000

100.104.0510

iHILIC®-Fusion(+) HILIC Column, PEEK, 100x4.6 mm, 5µm, 100Å

¥117,000

110.154.0510

iHILIC®-Fusion HILIC Column, PEEK, 150x4.6 mm, 5µm, 100Å

¥135,000

100.154.0510

iHILIC®-Fusion(+) HILIC Column, PEEK, 150x4.6 mm, 5µm, 100Å

¥135,000

110.254.0510

iHILIC®-Fusion HILIC Column, PEEK, 250x4.6 mm, 5µm, 100Å

¥169,000

100.254.0510

iHILIC®-Fusion(+) HILIC Column, PEEK, 250x4.6 mm, 5µm, 100Å

¥169,000

111.122.0510

iHILIC®-Fusion Guard Column,PEEK,20x2.1mm,5µm, 100Å, 1 pk

¥49,000

101.122.0510

iHILIC®-Fusion(+) Guard Column, PEEK, 20x2.1mm,5µm, 100Å, 1 pk

¥49,000

111.322.0510

iHILIC®-Fusion Guard Column,PEEK,20x2.1mm,5µm, 100Å, 3 pk

¥115,000

101.322.0510

iHILIC®-Fusion(+) Guard Column, PEEK, 20x2.1mm,5µm, 100Å, 3 pk

¥115,000P/N

iHILIC®-Fusion HPLC Column, SS – Stainless steel column hardware (SFC)

価格

P/N

iHILIC®-Fusion HPLC Column, SS – Stainless steel column hardware (SFC)

価格

114.053.0310

iHILIC®-Fusion HILIC Column, SS, 50x3.0 mm, 3.5µm, 100Å

¥120,000

104.053.0310

iHILIC®-Fusion(+) HILIC Column, SS, 50x3.0 mm, 3.5µm, 100Å

¥120,000

114.103.0310

iHILIC®-Fusion HILIC Column, SS, 100x3.0 mm, 3.5µm, 100Å

¥130,000

104.103.0310

iHILIC®-Fusion(+) HILIC Column, SS, 100x3.0 mm, 3.5µm, 100Å

¥130,000iHILIC®-Fusion(+) HILIC Column, SS, 150x3.0 mm, 3.5µm, 100Å

¥148,000

114.153.0310

iHILIC®-Fusion HILIC Column, SS, 150x3.0 mm, 3.5µm, 100Å

¥148,000

104.153.0310

114.253.0310

iHILIC®-Fusion HILIC Column, SS, 250x3.0 mm, 3.5µm, 100Å

¥169,000

104.253.0310

iHILIC®-Fusion(+) HILIC Column, SS, 250x3.0 mm, 3.5µm, 100Å

¥169,000

114.053.0510

iHILIC®-Fusion HILIC Column, SS, 50x3.0 mm, 5µm, 100Å

¥120,000

104.053.0510

iHILIC®-Fusion(+) HILIC Column, SS, 50x3.0 mm, 5µm, 100Å

¥120,000

114.103.0510

iHILIC®-Fusion HILIC Column, SS, 100x3.0 mm, 5µm, 100Å

¥130,000

104.103.0510

iHILIC®-Fusion(+) HILIC Column, SS, 100x3.0 mm, 5µm, 100Å

¥130,000¥143,000

104.153.0510

iHILIC®-Fusion(+) HILIC Column, SS, 150x3.0 mm, 5µm, 100Å

¥143,000¥164,000

104.253.0510

iHILIC®-Fusion(+) HILIC Column, SS, 250x3.0 mm, 5µm, 100Å

¥164,000

114.153.0510

iHILIC®-Fusion HILIC Column, SS, 150x3.0 mm, 5µm, 100Å

114.253.0510

iHILIC®-Fusion HILIC Column, SS, 250x3.0 mm, 5µm, 100ÅP/N

iHILIC®-Fusion(P) HPLC Column, PEEK – Engineering plastic column hardware

価格P/NiSPE®-HILIC SPE Cartridge for HILIC

130.052.0520 iHILIC®-Fusion(P) HILIC Column, PEEK, 50x2.1 mm, 5µm, 200Å

¥123,000

200-001-0025

iSPE-HILIC SPE Cartridge, 1 mL, 25 mg, 50 µm particle

130.102.0520 iHILIC®-Fusion(P) HILIC Column, PEEK, 100x2.1 mm, 5µm, 200Å

¥135,000

200-001-0050

iSPE-HILIC SPE Cartridge, 1 mL, 50 mg, 50 µm particle

100¥70,000

130.152.0520 iHILIC®-Fusion(P) HILIC Column, PEEK, 150x2.1 mm, 5µm, 200Å

¥148,000

200-001-0100

iSPE-HILIC SPE Cartridge, 1 mL, 100 mg, 50 µm particle

100¥84,000

130.054.0520 iHILIC®-Fusion(P) HILIC Column, PEEK, 50x4.6 mm, 5µm, 200Å

¥133,000

200-003-0100

iSPE-HILIC SPE Cartridge, 3 mL, 100 mg, 50 µm particle

50¥52,000

130.104.0520 iHILIC®-Fusion(P) HILIC Column, PEEK, 150x4.6 mm, 5µm, 200Å

¥151,000

200-003-0200

iSPE-HILIC SPE Cartridge, 3 mL, 200 mg, 50 µm particle

50¥61,000

130.154.0520 iHILIC®-Fusion(P) HILIC Column, PEEK, 150x4.6 mm, 5µm, 200Å

¥170,000

200-003-0500

iSPE-HILIC SPE Cartridge, 3 mL, 500 mg, 50 µm particle

50¥99,000

131.122.0520 iHILIC®-Fusion(P) HILIC Column, PEEK, 20x2.1 mm,5µm, 200Å, 1 pk

¥58,000

200-006-0500

iSPE-HILIC SPE Cartridge, 6 mL, 500 mg, 50 µm particle

25¥63,000

131.322.0520 iHILIC®-Fusion(P) HILIC Column, PEEK, 20x2.1 mm,5µm, 200Å, 3 pk

¥141,000

200-006-1000

iSPE-HILIC SPE Cartridge, 6 mL, 1 g, 50 µm particle

25¥99,000

P/N

iHILIC®-Fusion HPLC Column, SS – Stainless steel column hardware

価格※上記価格は2019年4月1日から適用されます。

134.053.0510

iHILIC®-Fusion(P) HILIC Column, SS, 50x3.0 mm, 5µm, 200Å

¥130,000※上記価格には消費税は含まれておりません。

134.103.0510

iHILIC®-Fusion(P) HILIC Column, SS, 100x3.0 mm, 5µm, 200Å

¥143,000

134.153.0510

iHILIC®-Fusion(P) HILIC Column, SS, 150x3.0 mm, 5µm, 200Å

¥156,000

輸入元

Units/Box

販売元

株式会社バイオニック

株式会社 クロマニックテクノロジーズ

417-0801 富士市大渕3397-19
TEL: 0545-38-9125 FAX: 0545-38-9126
Email: info@bionikinc.com URL: http:// www.bionikinc.com

552-0001 大阪府大阪市港区波除6-3-1
TEL: 06-6581-0885 FAX: 06-6581-0890
E-mail: info@chromanik.co.jp URL: http://chromanik.co.jp
首都圏オフィス 252-0331 相模原市南区大野台1-8-31-601
TEL: 042-851-3497 E-mail: sato@chromanik.co.jp

100

価格

¥66,000

1904