MCU SERIESELECTRIC TOTALIZING COUNTERNON-RESETTABLE
MODELSMODELS

TYPE

Additive
Total Counter

RESET

MCU-6SUL DC24V
MCU-6SUL AC100V
MCU-6SUL AC115V
MCU-6S AC200V
MCU-6S AC220V
MCU-6SUL DC24V B2
MCU-6SUL AC115V B2
MCU-7S DC24V
MCU-7S AC100V
MCU-7S AC200V

MODELS

Additive
Total Counter

Non-Reset

MODELS

RESET

Additive
Total Counter

Additive
Total Counter

RESET

Rear (Screw)

85 grams[3.00 oz]

Base (Screw | Vertical)

105 grams[3.70 oz]

Rear (Screw)

85 grams[3.00 oz]

DIGITSVOLTAGEMOUNTINGWEIGHT

DC24V
AC100V
AC115V
AC200V
AC220V
DC24V
AC100V
AC200V

Panel (Screw)

100 grams[3.53 oz]

VOLTAGEMOUNTING

WEIGHT

DC24V
AC100V
AC200V
AC220V
DC24V
AC100V
AC200V

Base(Screw | Horizontal)

100 grams[3.53 oz]

VOLTAGEMOUNTING

WEIGHT

DC24V
AC100V
AC200V
AC220V
DC24V
AC100V
AC200V

Base (Screw | Vertical)

105 grams[3.70 oz]

67

DIGITS
6Non-Reset
7

MODELSTYPE

WEIGHT

DC24V
AC100V
AC115V
AC200V
AC220V
DC24V
AC115V
DC24V
AC100V
AC200V7

MCU-6PUL DC24V
MCU-6PUL AC100V
MCU-6PUL AC115V
MCU-6P AC200V
MCU-6P AC220V
MCU-7P DC24V
MCU-7P AC100V
MCU-7P AC200V

MCU-6C DC24V
MCU-6C AC100V
MCU-6C AC200V
MCU-6C AC220V
MCU-7C DC24V
MCU-7C AC100V
MCU-7C AC200V

MOUNTING

VOLTAGE

6Non-Reset

TYPE

TYPE

DIGITSRESET

MCU-6CY DC24V
MCU-6CY AC100V
MCU-6CY AC200V
MCU-6CY AC220V
MCU-7CY DC24V
MCU-7CY AC100V
MCU-7CY AC200V

DIGITS
6Non-Reset
7SPECIFICATIONS
Display
Power Source
Voltage Fluctuation
Power Consumption
Continuous Energizing
Counting Method
Counting Speed
Pulse Width
Make (Duty)
ResetConnection
Operating Temperature
Operating Humidity
Test Voltage
Insulation ResistanceVibration
ShockConstruction

6 digits : 4.3 x 2.6mm [0.17 x 0.10 "] (white on black background) / 7 digits : 4.3 x 2.0 [0.17 x 0.08 "] (white on black background)AC 100V, 115V, 200V, 220V 50/60 Hz convertible / DC 24V±10%
AC : 3.2 - 4.3 VA; DC : 1.7W
AC : less than 5 minutes / DC : permissible
One pulse - One count (energizing - ½ count, unenergized - ½ count )10 Hz (maximum)
50msec. minimum
1:1NoneLead wire 250 mm [9.84 "]-5°C - +40°C [23 - 104 °F ] (non-freezing)
85%RH maximum (non-condensing)
1,500 VAC for 1 minute
20M minimum (500 VDC megger)
Endurance : 16.7 Hz, Width, double amplitude 4mm [0.16 "]Malfunction : 10 - 55 Hz, double amplitude 0.5mm [0.02 "]Endurance : 300 m/s² (30G)
Malfunction : 50 m/s² (5G)
Cover : Plastic (black)

DIMENSIONS - MILLIMETERS [INCH]

5.3[0.21]

6.2[0.24]

31[1.22]

MCU - 6P • 7P

3.3[0.13]

53.2[2.09]

1.6[0.06]

9[0.35] 2-M3

44[1.73] +0.5[0.02]
02- 3.6[0.14]hole

27[1.06] +0.5[0.02]
035.6[1.4]±0.2[0.01]
6.2[0.24]

22.2[0.87]
46[1.81]46[1.81]41.4[1.63]
18.1[0.71]
8.2[0.32]

1.6[0.06]

35.6[1.4]±0.2[0.01]

56.4[2.22]
11.5[0.45]
24.5[0.96]

41[1.61]

16[0.63]±0.2[0.01]
18.5[0.73]
24.8[0.98]

MCU - 6S • 7S

PANEL CUTOUT

2- 7.5[0.3]countersink

31[1.22] 7.3[0.29]

MCU - 6S B2
MCU - 6CY • 7CY

55.8[2.20]
47.8[1.88]±0.2[0.01]
41[1.61]

5[0.2]

56.3[2.22]

5[0.2]

31[1.22]4- 3.4[0.13]hole

24.5[0.96]

6.2[0.24]

11.5[0.45]

41.4[1.63]
41[1.61]
28.6[1.13]±0.2[0.01]

31[1.22]4[0.16]

7[0.28]60[2.36]47.2[1.86]±0.2[0.01]

56.3[2.22]
11.5[0.45]
24.5[0.96]

25.6[1.01]
17.6[0.69]±0.2[0.01]
6.2[0.24]

MCU - 6C • 7C

This catalog was last revised as of Jul. 13, 2023.
*Specifications and printed contents are subject to change without prior notice.
All Rights Reserved, Copyright © 2023,LINE SEIKI CO.,LTD.

LINE SEIKI CO., LTD.
2-37-7 Chuo-cho,Meguro-ku, Tokyo 152-0001 Japan
Please visit our company website for contact details
webtrade@line.co.jp
https://www.lineseiki.com