Instructions for Use

Color LCD Monitor

ImportantPlease make sure you read the Instructions for Use and Installation
Manual before use.
Please retain this manual for future reference.
• See the Installation Manual for monitor settings and adjustments.
• The latest product information, including the Instructions for Use, isavailable on our web site.
www.eizoglobal.com

SAFETY SYMBOLSThis manual and this product use the safety symbols below. They denote criticalinformation. Please read them carefully.
Failure to abide by the information in a WARNING may result in serious injury
and can be life threatening.

WARNING
Failure to abide by the information in a CAUTION may result in moderate injury
and/or property or product damage.

CAUTION
Indicates a warning or caution. For example,
shock" hazard.

indicates an "electrical

Indicates a prohibited action. For example,

means "Do not disassemble".

This product has been adjusted specifically for use in the region to which it was originallyshipped.
If operated outside this region, the product may not perform as stated in the specifications.
No part of this manual may be reproduced, stored in a retrieval system, or transmitted, in
any form or by any means, electronic, mechanical, or otherwise, without the prior written
permission of EIZO Corporation.
EIZO Corporation is under no obligation to hold any submitted material or information
confidential unless prior arrangements are made pursuant to EIZO Corporation’s receipt ofsaid information. Although every effort has been made to ensure that this manual provides
up-to-date information, please note that EIZO product specifications are subject to change
without notice.
2

PRECAUTIONSPRECAUTIONS
Important
This product has been adjusted specifically for use in the region to which it was originallyshipped. If operated outside this region, the product may not perform as stated in thespecifications.
To personal safety and proper maintenance, please read carefully the "PRECAUTIONS"
section and the caution statements on the monitor.

Location of Caution Statement

Symbols on the unit
SymbolThis symbol indicates
Main power switch: Press to turn the monitor’s main power off.
Main power switch: Press to turn the monitor’s main power on.
Power button: Press to turn the monitor’s power on or off.
Alternating current
Alerting electrical shock hazard
CAUTION: See SAFETY SYMBOLS [} 2]
WEEE marking: Product must be disposed of separately; materials may berecycled.
CE marking: EU conformity mark in accordance with the provisions of Council
Directive and/or Regulation (EU).
Manufacturer
Date of manufacture

3

PRECAUTIONS

SymbolThis symbol indicates
Caution: Federal law (USA) restricts this device to sale by or on the order of alicensed healthcare practitioner.
Medical device in the EU
Importer in the EU
Mark signifying compliance with UK regulations
UK Responsible Person
Authorised representative in Switzerland
Authorised representative in the European Community

4

PRECAUTIONSWARNINGWARNING
If the unit begins to emit smoke, smells like something is burning, or makes strange noises,
disconnect all power connections immediately and contact your EIZO representative foradvice.
Attempting to use a malfunctioning unit may result in fire, electrical shock, or equipment damage.

WARNING
Do not disassemble or modify the unit.
Opening the cabinet may result in electrical shock or burn by high-voltage or hightemperature parts. Modifying the unit may result in fire or electrical shock.

WARNING
Refer all servicing to qualified service personnel.
Do not attempt to service this product yourself as opening or removing covers may result in fire,electrical shock, or equipment damage.

WARNING
Keep foreign objects or liquids away from the unit.
Metal parts, flammable materials, or liquids accidentally falling into the cabinet may
result in fire, electrical shock, or equipment damage.
If an object or liquid falls/spills into the cabinet, unplug the unit immediately. Have theunit checked by a qualified service engineer before using it again.

WARNING
Place the unit at a sturdy and stable place.
A unit placed on an inadequate surface may fall and result in injury.
If the unit falls, disconnect the power immediately and ask your local EIZO representative for advice.
Do not continue using a damaged unit. Using a damaged unit may result in fire or electrical shock.

5

PRECAUTIONSWARNING
Use the unit in an appropriate location.
Otherwise, fire, electrical shock, or equipment damage may result.
• Do not place outdoors.
• Do not place in any form of transportation (ships, aircraft, trains, automobiles, etc.).
• Do not place in dusty or humid environments.
• Do not place in locations where water may be splashed on the screen (bathrooms,
kitchens, etc.)• Do not place in locations where steam comes in direct contact with the screen.
• Do not place near heat generating devices or humidifiers.
• Do not place in locations where the product is subject to direct sunlight.
• Do not place in environments with inflammable gas.
• Do not place in environments with corrosive gases (such as sulfur dioxide,hydrogen sulfide, nitrogen dioxide, chlorine, ammonia, and ozone).
• Do not place in environments with dust, components that accelerate corrosion in
the atmosphere (such as sodium chloride and sulfur), conductive metals, and soon.

WARNING
Keep the plastic packing bags away from babies and children.
The plastic packing bags may cause suffocation.

WARNING
Use the enclosed power cord and connect to the standard power outlet in your country.
Be sure to use within the rated voltage of the power cord. Otherwise, fire or electric shock may
result.
Power supply: 100-240 Vac 50/60 Hz

WARNINGTo disconnect the power cord, grasp the plug firmly and pull.
Tugging on the cord may damage and result in fire or electrical shock.

WARNINGThe equipment must be connected to a grounded main outlet.
Failure to do so may result in fire or electric shock.

6

PRECAUTIONSWARNING
Use the correct voltage.
• The unit is designed for use with a specific voltage only. Connection to another voltage thanspecified in this "Instructions for Use" may cause fire, electric shock, or equipment damage.
Power supply: 100-240 Vac 50/60 Hz• Do not overload your power circuit, as this may result in fire or electric shock.

WARNING
Handle the power cord with care.
Do not place heavy objects on the power cord, or pull or tie the power cord. Using adamaged power cord may result in fire or electrical shock.

WARNINGThe operator should not touch the patient while touching the product.
This product has not been designed to be touched by patients.

WARNING
Never touch the plug and power cord if it begins to thunder.
Touching them may result in electrical shock.

WARNING
When attaching an arm stand, please refer to the User’s Manual of the arm stand and install
the unit securely.
Otherwise, the unit may become detached, resulting in injury and/or equipment damage.
Before installation, make sure that desks, walls, or any other installation surface has adequate
mechanical strength.
If the unit falls, disconnect the power immediately and ask your local EIZO representative for advice.
Do not continue using a damaged unit. Using a damaged unit may result in fire or electrical shock.
When reattaching the tilt stand, please use the same screws and tighten them securely.

WARNING
Do not touch a damaged LCD panel directly with bare hands.
If any part of your skin comes in direct contact with the panel, wash thoroughly.
If liquid crystal enters your eyes or mouth, immediately flush with large amounts ofwater and seek medical attention. Otherwise, you may have a toxic reaction.

WARNINGFor installment in high locations, ask for the help of a professional.
When installing the monitor in a high location, there is a risk of the product or its parts falling andcausing injury. Ask for help from us or from a professional who specializes in construction workwhen installing the monitor, including an inspection of the product for any damage or deformities
both before and after installing the monitor.

7

PRECAUTIONSCAUTIONCAUTION
Check the operational state before use.
Begin use after ensuring that there are no problems with the displayed image.
When using multiple units, begin use after ensuring that the images are displayed appropriately.

CAUTION
Securely fix cables / cords that have a fixing feature.
If they are not fixed securely, cables / cords may disconnect, and subsequently images may be cut
off and your operations may be disrupted.

CAUTION
Disconnect cables and remove accessories when moving the unit.
Otherwise, the cables or accessories may become detached when moving, resulting in injury.

CAUTION
Carry or place the unit according to the correct specified methods.
• When moving the product, hold the bottom of the monitor firmly.
• Monitors of size 30 inches and above are heavy. When unpacking and/or carrying
the monitor, ensure at least two people are utilized.
• If your device model has a handle on the back of the monitor, grasp and firmly holdthe bottom and handle of the monitor.
Dropping the unit may result in injury or equipment damage.

CAUTION
Exercise care not to pinch your hands.
If you suddenly apply force to the monitor to adjust its height or angle, your hands may be pinched
and injured.

CAUTION
Do not block the ventilation slots on the cabinet.
• Do not place any objects on the ventilation slots.
• Do not install the unit in a place with poor ventilation or inadequate space.
• Do not use the unit laid down or upside down.
Blocking the ventilation slots prevents proper airflow and may result in fire, electricalshock, or equipment damage.

8

PRECAUTIONSCAUTION
Do not touch the power plug with wet hands.
Touching them may result in electrical shock.

CAUTION
Do not place any objects around the power plug.
This is to facilitate disconnecting the power plug in case of a problem to avoid fire orelectrical shock.

CAUTION
Periodically clean the area around the power plug and the ventilation slot of the monitor.
Dust, water, or oil adhered to this area may result in fire.

CAUTION
Unplug the unit before cleaning it.
Cleaning the unit while it is plugged into a power outlet may result in electrical shock.

CAUTION
If you plan to leave the unit unused for an extended period of time, disconnect the powerplug from the wall socket after turning off the power switch for the safety and the powersaving.

CAUTION
Dispose of this product in accordance with the laws of the locality or country of residence.

CAUTIONFor users in the territory of the EEA and Switzerland:
Any serious incident that has occurred in relation to the device should be reported to the
Manufacturer and the Competent Authority of the Member State in which the user and/or
patient is established.

9Notice for this monitor

Notice for this monitor
Indications for Use
This Product is indicated for use in displaying radiological images for review, analysis, anddiagnosis by trained medical practitioners. The display is not intended for mammography.
Attention• This product may not be covered by warranty for uses other than those described in this manual.
• The specifications noted in this manual are only applicable when using the power cords provided
together with the product and EIZO-specified signal cables.
• Only use EIZO accessories products specified by EIZO with this product.

Precautions for Use• Parts (such as the LCD panel and the fan) may deteriorate if used over extended
periods of time. Periodically check that they are operating normally.
• When the screen image is changed after displaying the same image for extended
periods of time, an afterimage may remain. Use the screensaver or power save function
to avoid displaying the same image for extended periods of time. An afterimage may
appear even after a short time period has elapsed depending on the displayed image.
To remove such a phenomenon, change the image or keep the power turned off forseveral hours.
• It takes about several minutes for the monitor display to stabilize. Before using themonitor, wait a few minutes or longer after the monitor power has been turned on orafter the monitor has recovered from the power saving mode.
• If the monitor displays continuously over a long period of time, smudges or burn-in may
appear. To maximize the life of the monitor, we recommend the monitor be turned off
periodically.
• The backlight of the LCD panel has a fixed lifetime. Depending on the usage pattern,such as usage for long continuous periods, the lifespan of the backlight may run out
sooner, requiring replacement. When the screen becomes dark or begins to flicker,
please contact your local EIZO representative.
• The screen may have defective pixels or a small number of light dots on the screen.
This is due to the characteristics of the LCD panel itself, and is not a malfunction of the
product.
• Do not press on the LCD panel surface or edge of the frame strongly, as this may resultin display malfunctions, such as moiré patterns, etc. If pressure is continuously applied
to the LCD panel surface, the liquid crystal may deteriorate or the LCD panel may bedamaged. (If the pressure marks remain on the LCD panel, leave the monitor with ablack or white screen. The symptom may disappear.)• Do not scratch or press on the LCD panel with any sharp objects, at this may result indamage to the LCD panel. Do not attempt to brush with tissues as this may scratch thepanel.
• Do not touch the built-in calibration sensor (Integrated Front Sensor). Doing so may
reduce the measurement accuracy or result in equipment damage.
• Depending on the environment, the value measured by the built-in illuminance sensor
may differ from the value shown on a stand-alone illuminometer.
• Dew condensation may form on the surface or interior of this product when it is brought
into a cold room, when the temperature suddenly rises, or when it is moved from a cold10

Notice for this monitor
room to a warm room. In that case, do not turn the product on. Instead wait until the dew
condensation disappears, otherwise it may cause some damage to the product.

To Use the Monitor for a Long Time
Quality control• The display quality of monitors is affected by the quality level of input signals and thedegradation of the product. Perform visual checks and periodic constancy tests(including grayscale check) to comply with medical standards / guidelines according to
your application, and carry out calibration as necessary. The RadiCS (optional) monitor
quality control software enables you to perform quality control that meets medical
standards / guidelines.
• Please wait 15 minutes or more after the monitor power has been turned on or themonitor has recovered from the power saving mode before performing various tests for
quality control, calibration, or screen adjustment of the monitor.
• We recommend that monitors be set to the recommended level or lower to reducechanges in luminosity caused by long-term use and maintain stable brightness.
• To adjust measurement results of the integrated calibration sensor (Integrated Front
Sensor) to those of an EIZO external sensor (UX2 sensor) that is sold separately,
perform correlation between the Integrated Front Sensor and the external sensor using
RadiCS (optional accessory) / RadiCS LE (included). Periodical correlation allows you tomaintain the measurement result of the Integrated Front Sensor at a level equivalent tothat of the external sensor. The accuracy of the Integrated Front Sensor of this productis optimized for the portrait display in its factory setting. When performing quality control
using the Integrated Front Sensor in a landscape display, be sure to perform correlationin the landscape display. For details on correlation, refer to the RadiCS / RadiCS LE
User's Manual.
Attention• The display status of the monitor may change unexpectedly due to an operating error or
unexpected setting change. Using the monitor with the operation switches locked isrecommended after adjusting the screen of the monitor.
For details on how to set, refer to the Installation Manual (on the CD-ROM).

Cleaning• Periodic cleaning is recommended to keep the product looking new and to prolong its
operation lifetime.
• Stains on the product can be removed by moistening part of a soft cloth with water or byusing our ScreenCleaner and gently wiping.
Attention• Do not let liquids come into direct contact with the product. If it does, wipe it off immediately.
• Do not allow liquids to get into gaps or inside the product.
• When using chemicals for cleaning or disinfection, chemicals such as alcohol and disinfectant
may cause gloss variation, tarnishing, and fading of the product, and also quality deterioration of
the displayed image. Do not use chemicals on a frequent basis.
• Never use a thinner, benzene, wax, or abrasive cleaner as they may damage the product.
• For more information on cleaning and disinfection, please refer to our web site.
How to check: Access www.eizoglobal.com and type "disinfect" in the site search box to search.

11

Notice for this monitor

Disinfection with chemicals• When disinfecting products, we recommend using chemicals which have been tested byus (see table below). Note that using these chemicals do not guarantee that the productwill not be damaged or deteriorated.
Category

Chemical type

Product example

Alcohol-based

Rubbing ethanol (ethyl alcohol)

Ethanol

Alcohol-based

Isopropanol

Isopropyl alcohol (IPA)

Chlorine

Sodium hypochlorite

PureloxBiguanide

Chlorhexidine gluconate

Hibitane solution

Alcohol-based

Benzalkonium chloride

Welpas

Aldehyde-based

Glutaral

Sterihyde

Aldehyde-based

GlutaralCidex Plus28

Amphoteric surfactant

Alkyldiaminoethylglycine hydrochloride

Satenidin solution

To Use the Monitor Comfortably• An excessively dark or bright screen may affect your eyes. Adjust the brightness of themonitor according to the environmental conditions.
• Staring at the monitor for a long time tires your eyes. Take a 10-minute rest every hour.
• Look at the screen from a proper distance and from a proper angle.

Cybersecurity Warnings and Responsibilities• Firmware update should be performed through EIZO Corporation or its distributor.
• If EIZO Corporation or its distributor instructs to update the firmware, update it
immediately.

12

CONTENTS

CONTENTS
PRECAUTIONS ...........................................................................................................

3

Important .............................................................................................................

3

Location of Caution Statement................................................................

3

Symbols on the unit ................................................................................

3

Notice for this monitor............................................................................................... 10
Indications for Use .............................................................................................. 10
Precautions for Use............................................................................................. 10
To Use the Monitor for a Long Time ................................................................... 11
Quality control ......................................................................................... 11
Cleaning.................................................................................................. 11
Disinfection with chemicals ..................................................................... 12
To Use the Monitor Comfortably ......................................................................... 12
Cybersecurity Warnings and Responsibilities ..................................................... 12
1 Introduction ................................................................................................................ 15
1.1

1.2

1.3

Features .............................................................................................................. 15
1.1.1

Hybrid monochrome and color display.................................................... 15

1.1.2

Simple wiring........................................................................................... 15

1.1.3

Quality control ......................................................................................... 15

1.1.4

Monitor operation from the mouse and keyboard ................................... 16

Package Contents............................................................................................... 16
1.2.1

EIZO LCD Utility Disk.............................................................................. 16

1.2.2

RadiCS LE .............................................................................................. 16

1.2.3

To use RadiCS LE .................................................................................. 17

Controls and Functions ....................................................................................... 18
1.3.1

Front........................................................................................................ 18

1.3.2

Back ........................................................................................................ 19

2 Installation / Connection............................................................................................ 20
2.1

Before installation................................................................................................ 20
2.1.1

Installation conditions.............................................................................. 20

2.2

Connecting Cables.............................................................................................. 21

2.3

Turning On the Power ......................................................................................... 22

2.4

Adjusting the Screen Height and Angle .............................................................. 23

3 No-Picture Problem.................................................................................................... 24
4 Specifications ............................................................................................................. 26
4.1

List of Specifications ........................................................................................... 26
4.1.1

LCD Panel............................................................................................... 26

4.1.2

Video Signals .......................................................................................... 26

4.1.3

USB......................................................................................................... 26
13

CONTENTS

4.1.4

Power...................................................................................................... 27

4.1.5

Physical Specifications ........................................................................... 27

4.1.6

Operating Environmental Requirements................................................. 27

4.1.7

Transport / Storage Conditions ............................................................... 27

4.2

Compatible Resolutions ...................................................................................... 28

4.3

Accessories......................................................................................................... 28

Appendix ..................................................................................................................... 29
Medical Standard ................................................................................................ 29
Classification of Equipment..................................................................... 29
EMC Information ................................................................................................. 30
Environments of Intended Use................................................................ 30
Technical Descriptions............................................................................ 31
Information for Radio Interference ...................................................................... 37
For U.S.A, Canada Only ......................................................................... 37

14

Introduction | 1

1 Introduction
Thank you very much for choosing an EIZO color LCD monitor.

1.1 Features
1.1.1 Hybrid monochrome and color display
When the Hybrid Gamma PXL function is enabled, this product automatically differentiates
between monochrome and color parts of the same image at a pixel level and displays them
respectively in optimal gradients.

1.1.2 Simple wiring
In addition to a DisplayPort input terminal, an output terminal is also provided.
Signals can be output from the DisplayPort output connector to other monitors.

1.1.3 Quality control• This monitor has a built-in calibration sensor (Integrated Front Sensor). This sensor
enables the monitor to perform calibration (SelfCalibration) and Grayscale Check
independently.

• Using the included RadiCS LE allows you to perform monitor calibration and history
management.
• The RadiCS monitor quality control software enables you to perform quality control thatmeets medical standards / guidelines.

15

1 | Introduction

1.1.4 Monitor operation from the mouse and keyboard
Using the RadiCS / RadiCS LE monitor quality control software, you can perform thefollowing monitor operations using the mouse and keyboard:• Switching CAL Switch modes• Switching input signals• Function that assigns any CAL Switch mode to a part of the screen and displays animage (Point-and-Focus)• Entering power saving mode (Backlight Saver)

1.2 Package Contents
Check that all of the following items are contained in the package. If any of these are
missing or damaged, contact your dealer or local EIZO representative.
Note• It is recommended that the box and packing materials be kept in storage so that they can beused to move or transport this product later on.

• Monitor• Power cord

• Digital signal cable (DisplayPort - DisplayPort): PP300 x 1• USB cable: UU300 x 1• EIZO LCD Utility Disk (CD-ROM)• Instructions for Use

1.2.1 EIZO LCD Utility Disk
The CD-ROM contains the following items. Refer to "Readme.txt" on the disk for software
startup procedures or file reference procedures.
• Readme.txt file
• RadiCS LE monitor quality control software (for Windows)• User’s Manual– Instructions for Use of this monitor– Monitor Installation Manual– RadiCS LE User’s Manual• Outline dimensions

1.2.2 RadiCS LE
RadiCS LE enables you to perform the following quality control and monitor operations. For
more information about the software or setup procedures, refer to RadiCS LE User’s
Manual.
Quality control• Executing calibration
16

Introduction | 1• Displaying test results in a list and creating a test report• Setting the SelfCalibration target and execution schedule
Monitor operations• Switching CAL Switch modes• Switching input signals• Function that assigns any CAL Switch mode to a part of the screen and displays animage (Point-and-Focus)• Entering power saving mode (Backlight Saver)• A feature that automatically adjusts monitor brightness to match the ambient light level
when set to Text mode (Auto Brightness Control)
Attention• The specifications of RadiCS LE are subject to change without notice. The latest version of
RadiCS LE is available for download from our web site: (www.eizoglobal.com)

1.2.3 To use RadiCS LE
For information on how to install and use RadiCS LE, refer to RadiCS LE User’s Manual (onthe CD-ROM).
When using RadiCS LE, connect the monitor to your PC using the supplied USB cable. For
more information about how to connect the monitor, see 2.2 Connecting Cables [} 21].

17

1 | Introduction

1.3 Controls and Functions
1.3.1 Front
1

2

Approx. 30°

34

1. Integrated Front Sensor(Movable)This sensor is used to perform calibration and Grayscale Check.

2. Illuminance Sensor

This sensor measures environmental illumination. Environmental
illumination measurement is performed using the RadiCS / RadiCS
LE quality control software.

3. Operation switches

Displays the operation guide. Set menus according to the operation
guide.

4.

Turns the power on or off.

switchThe switch indicator is lit when you turn the power on. The indicator
color differs depending on the monitor's operation status.
Green: Normal operation mode, Orange: Power saving mode,
Off: Main power or power turned off

18

Introduction | 1

1.3.2 Back
1

3

2

5

4
6

7

1. Handle

8

9

10

11

This handle is used for transportation.
Attention• Firmly hold the monitor by the handle and the bottom while
carrying it, and do not put pressure on the LCD panel or drop
the monitor. Do not hold the sensor section on the front side of
the monitor.

2. Stand

Adjusts the height and angle (tilt, swivel. rotation) of the monitor.

3. Main power switch

Turns the main power on or off.
: Off,: On

4. Security lock slot

Complies with Kensington's MicroSaver security system.

5. Cable holder

Holds the monitor cables.

6. USB Type-A connector(USB downstream port)

Connect it to a USB device. To set up a daisy-chain connection,connect the cable to the USB upstream port of another monitor.

7. USB Type-B connector(USB upstream port)

Connect this port to the PC when you use software that needs a
USB connection or you connect a USB device (a peripheral device
that supports USB) to the USB downstream port.

8. DVI-D connector

Connect it to the PC.

9. DisplayPort input
connector
10. DisplayPort output
connector

To set up a daisy-chain connection, connect the cable to the
DisplayPort input connector of another monitor.

11. Power connector

Connects the power cord.

19

2 | Installation / Connection

2 Installation / Connection
2.1 Before installation
Carefully read PRECAUTIONS [} 3] and always follow the instructions.
If you place this product on a lacquer-coated desk, the color may adhere to the bottom of
the stand due to the composition of the rubber. Check the desk surface before use.

2.1.1 Installation conditions
When installing the monitor in a rack, ensure that there is adequate space around the sides,back and top of the monitor.
Attention• Position the monitor so that there is no light to interfere with the screen.

20Installation / Connection | 2

2.2 Connecting Cables
Attention• Check that the monitor, PC, and peripherals are powered off.
• When replacing the current monitor with this monitor, see 4.2 Compatible Resolutions [} 28] tochange the PC settings for resolution and vertical scan frequency to those that are available forthis monitor, before connecting the PC.

1. Connect signal cables.
Check the shapes of the connectors, and connect the cables. After connecting the DVI
cable, tighten the fasteners to secure the connector.

Signal cable: DD300 / DD200(optional)or
Signal cable: PP300

Attention• The DisplayPort connectors of the monitor consist of input and output connectors. Whenconnecting the monitor to a PC, connect the cable to the input connector.
• When connecting to multiple PCs, switch the input signal. For details, refer to the Installation
Manual (on the CD-ROM).

Note• If the cables are difficult to insert, adjust the angle of the screen.

When connecting other monitors using a daisy-chain connection
The signal input to the DisplayPort input connector can be output to another monitor.

21

2 | Installation / Connection

Attention• Visit the EIZO web site for information about monitors and graphics boards that can be used forthe daisy-chain connection:(www.eizoglobal.com)• To set up a daisy-chain connection, make sure that "DisplayPort" in the Administrator Settings
menu is set to "Version 1.2". For details, refer to the Installation Manual (on the CD-ROM).
• Remove the DisplayPort output connector cap before connecting the signal cable.

Connect to PC
Signal cable: PP300

Signal cable: PP300

2. Plug the power cord into a power outlet and the power connector on the monitor.
Insert the power cord fully into the monitor.
3. Connect the USB cable to the monitor's USB upstream port and to the PC's USB
downstream port respectively.
The cable connection is required when you use RadiCS / RadiCS LE or when you
connect a USB device (USB-connected peripheral device) to the monitor.

USB cable: UU300

2.3 Turning On the Powerto turn on the power to the monitor.
1. Touch
The power switch indicator of the monitor lights up green.
If the indicator does not light up, see 3 No-Picture Problem [} 24].
Note
• When you touch any of the operation switches excluding
with the monitor power turned off,
starts flashing to let you know where the power switch is located.

22

Installation / Connection | 2
2. Turn on the PC.
The screen image appears.
If an image does not appear, see 3 No-Picture Problem [} 24] for additional advice.
Attention• When connecting for the first time or changing the connection method, display settings such as
resolution and display scale may not be appropriate. Check if the settings for the PC are properly
configured.
• For power saving purposes, it is recommended that the Power button be turned off. When notusing the monitor, you can turn off the main power supply or disconnect the power plug so thatthe power is cut completely.

Note• In order to maximize the monitor's lifespan by impeding brightness degradation and to reduce
power consumption, carry out the following:– Use the power saving function of the PC or monitor.
– Turn off the monitor after using it.

2.4 Adjusting the Screen Height and Angle
Hold the top and bottom or left and right edges of the monitor with both hands, and adjust
the screen height, tilt and swivel the screen to the optimum position for performing tasks.

Attention• After the adjustment is finished, make sure that the cables are correctly connected.
• After adjusting the height and angle, pass the cables through the cable holder.

23

3 | No-Picture Problem

3 No-Picture Problem
Power switch indicator does not light up• Check whether the power cord is connected properly.
• Turn the main power switch on.
• Touch.

• Turn off the main power, and then turn it on again.

Power switch indicator lights up: Green• Increase "Brightness", "Contrast", or "Gain" in the Setting menu (cannot be adjusted in
1-DICOM, 2-CAL1, 3-CAL2 modes). For details, refer to the Installation Manual (on the
CD-ROM).
• Turn off the main power, and then turn it on again.

Power switch indicator lights up: Orange• Switch the input signal. For details, refer to the Installation Manual (on the CD-ROM).
• Move the mouse or press any key on the keyboard.
• Check whether the PC is turned on.
• Check whether the signal cable is connected properly. Connect to the DisplayPort input
connector when inputting a DisplayPort signal. The DisplayPort output connector is used
for output when a daisy-chain connection is set up.
• Turn off the main power, and then turn it on again.

Power switch indicator blinks: Orange, Green• Connect via the signal cable specified by EIZO. Turn off the main power, and then turn it
on again.
• If the signal cable is connected to DisplayPort, try switching the DisplayPort version. For
details, refer to the Installation Manual (on the CD-ROM).

The message "No Signal" appears in the screen.
Example:• The message shown above may appear, because some PCs do not output the signal
immediately after power-on.
• Check whether the PC is turned on.
• Check whether the signal cable is connected properly. Connect the signal cables to theconnectors of the corresponding input signal.
• Switch the input signal. For details, refer to the Installation Manual (on the CD-ROM).
• If the signal cable is connected to DisplayPort, try switching the DisplayPort version. For
details, refer to the Installation Manual (on the CD-ROM).

24

No-Picture Problem | 3• Check whether the signal cable is connected properly. Check whether the signal cable
is connected to the DisplayPort input connector when inputting a DisplayPort signal. The
DisplayPort output connector is used for output when a daisy-chain connection is set up.
• Turn off the main power, and then turn it on again.

The message "Signal Error" appears on the screen
The message "Signal Error" appears in the screen.
Example:• Check whether the PC is configured to meet the resolution and vertical scan frequency
requirements of the monitor (see 4.2 Compatible Resolutions [} 28]).
• Reboot the PC.
• Select an appropriate setting using the graphics board’s utility. Refer to the User’s
Manual of the graphics board for details.
• If the signal cable is connected to DisplayPort, try switching the DisplayPort version. For
details, refer to the Installation Manual (on the CD-ROM).

25

4 | Specifications

4 Specifications
4.1 List of Specifications
4.1.1 LCD PanelType

IPS (Anti-Glare)

Backlight

LED

Size

21.3" (54.0 cm)

Resolution

1200 dots x 1600 lines

Display Size (H x V)

324.0 mm x 432.0 mm

Pixel Pitch

0.270 mm x 0.270 mm

Display Colors

10-bit color (DisplayPort): Up to 1.07 billion colors(from a palette of approx. 543 billion colors)
8-bit (DisplayPort / DVI): 16.77 million colors (froma palette of approx. 543 billion colors)

Viewing Angle (H / V, typical)

178˚ / 178˚

Recommended Brightness

270 cd/m2

Contrast Ratio (typical)

1800:1

Response Time (typical)

20 ms (black -> white -> black)4.1.2 Video Signals
Input Terminals

DisplayPort x 1, DVI-D (Single Link) x 1

Output Terminals

DisplayPort x 1

Horizontal scan frequency

31 kHz – 100 kHz

Vertical scan frequency*1

59 Hz – 61 Hz (720 x 400: 69 Hz – 71 Hz)

Frame Synchronization mode

59 Hz – 61 Hz

Dot clock

DisplayPort

25 MHz – 164.5 MHz

DVI

25 MHz – 164.5 MHz*1 The vertical scan frequency supported varies according to the resolution. For more information,

see 4.2 Compatible Resolutions [} 28].

4.1.3 USB
Port

Upstream

USB-B x 1

Downstream

USB-A x 2Standard
Supply Current

26

USB Specification Revision 2.0
Downstream (USB-A)

Max. 500 mA per port

Specifications | 4

4.1.4 PowerInput

100 – 240 VAC ±10 %, 50 / 60 Hz 0.85 – 0.45 A

Maximum Power Consumption

54 W or lessPower Save Mode

0.6 W or less*1

Standby Mode

0.6 W or less*2*1 When using DisplayPort import and the USB upstream port is not connected, "DP Power Save" :"On", "DisplayPort", "Version 1.1 10bit", "Version 1.1 8bit", no external load is connected*2 When the USB upstream port is not connected, "DP Power Save" : "On", "DisplayPort", "Version

1.1 10bit", "Version 1.1 8bit", no external load is connected

4.1.5 Physical Specifications
Dimensions (W x H x D)

356.5 mm x 482.3 mm – 572.3 mm x 200.0 mm(Tilt: 0˚)
356.5 mm x 506.1 mm – 596.1 mm x 260.9 mm(Tilt: 30˚)Dimensions (W x H x D) (Without Stand)356.5 mm x 464.5 mm x 70.5 mm

Net Weight

Approx. 7.2 kg

Net Weight (Without Stand)Approx. 4.3 kg

Height Adjustment Range

90 mm (Tilt: 0˚)

Tilt

Up 30˚, down 5˚Swivel70˚

Rotation

90˚ (Counterclockwise)

4.1.6 Operating Environmental Requirements
Temperature

0˚C – 35˚C

Humidity

20 % – 80 % R.H. (no dew condensation)

Air Pressure

540 hPa – 1060 hPa

4.1.7 Transport / Storage Conditions
Temperature-20˚C – 60˚C

Humidity

10 % – 90 % R.H. (no dew condensation)

Air Pressure

200 hPa – 1060 hPa

27

4 | Specifications

4.2 Compatible Resolutions
The monitor supports the following resolutions.
✓: Supported, -: Not supported
Resolution

Vertical scanfrequency (Hz)DisplayPort

DVI

Portrait

LandscapePortrait

Landscape720 x 400

70.087✓✓✓✓640 x 480

59.940✓✓✓✓800 x 600

60.317✓✓✓✓1024 x 768

60.004✓✓✓✓1280 x 1024

60.020

-

✓

-

✓

*1

60.000-✓-✓1200 x 1600*1

59.963✓-✓-1600 x 1200

*1 Recommended resolution

4.3 Accessories
The following accessories are available separately.
For the latest information about the optional accessories and information about the latest
compatible graphics board, refer to our web site.
(www.eizoglobal.com)
Calibration Kit

RadiCS UX2 Ver.5.1.1 or later
RadiCS Version Up Kit Ver.5.1.1 or later

Network QC Management Software

RadiNET Pro Ver.5.1.1 or later

Cleaning Kit

ScreenCleaner

Comfort Light for Reading Rooms

RadiLight

Arm

AAH-02B3W
LA-011-W

Panel Protector

RP-915
FP-2101

VESA Adapter for Thin Client or Mini-PC

PCSK-R1

Signal Cable (DVI-D - DVI-D)

DD300
DD200

28

Appendix

Appendix
Medical Standard• It shall be assured that the final system is in compliance to IEC60601-1 requirement.
• Power supplied equipment can emit electromagnetic waves, that could influence, limit or
result in malfunction of the monitor. Install the equipment in a controlled environment,
where such effects are avoided.

Classification of Equipment• Type of protection against electric shock : Class I• EMC class: IEC60601-1-2 Group 1 Class B• Classification of medical device (EU): Class I• Mode of operation : Continuous• IP Class : IPX029

Appendix

EMC Information
The RadiForce series has the capability to properly display medical images.

Environments of Intended Use
The RadiForce series is intended for use in the following environments• Professional healthcare facility environments such as clinics and hospitals
The following environments are not suitable for the RadiForce series to be used:• Home healthcare environments• In the vicinity of high-frequency surgical equipments such as electrosurgical knives• In the vicinity of short-wave therapy equipments• RF shielded room of the medical equipment systems for MRI• In shielded location Special environments• Installed in vehicles including ambulances• Other special environments
WARNING• The RadiForce series requires special precautions regarding EMC and need to be installed. You
need to carefully read EMC Information and the "PRECAUTIONS" section in this document, andobserve the following instructions when installing and operating the product.

WARNING• The RadiForce series should not be used adjacent to or stacked with other equipment. If
adjacent or stacked use is necessary, the equipment or system should be observed to verify
normal operation in the configuration in which it will be used.

WARNING• When using a portable RF communication equipment, keep it 30 cm (12 inches) or more awayfrom any part, including cables, of the RadiForce series. Otherwise, degradation of theperformance of this equipment could result.

WARNING• Anyone who connects additional equipment to the signal input part or signal output parts,
configuring a medical system, is responsible that the system complies with the requirements of
IEC60601-1-2.

WARNING• Do not touch the signal input/output connectors while using the RadiForce series. Otherwise, thedisplayed image may be affected.

30Appendix

WARNING• Be sure to use the cables attached to the product, or cables recommended by EIZO.
Use of cables other than those recommended by EIZO for this equipment could result inincreased electromagnetic emissions or decreased electromagnetic immunity of this equipment
and improper operation.
Cable

EIZO Designated
CablesMax. Cable
LengthShielding

Ferrite Core

Signal cable(DisplayPort)

PP300 / PP200

3m

ShieldedWithout Ferrite
Cores

Signal cable (DVI)

DD300 / DD200

3m

ShieldedWith Ferrite Cores

USB cable

UU300 / MD-C93

3m

ShieldedWith Ferrite Cores

Power cable (withgrounding wire)-

3m

Unshielded

Without Ferrite
Cores

Technical Descriptions
Electromagnetic emissions
The RadiForce series is intended for use in the electromagnetic environments specified
below.
The customer or the user of the RadiForce series should ensure that it is used in such anenvironment.
Emission test
RF emissions
CISPR11

ComplianceGroup 1

Electromagnetic environment - Guidance
The RadiForce series uses RF energy only for its internalfunction.
Therefore, its RF emission are very low and are not likely tocause any interference in nearby electronic equipment.

RF emissions
CISPR11Class B

Harmonic emissions
IEC61000-3-2Class D

Voltage fluctuations /flicker emissions
IEC61000-3-3Complies

The RadiForce series is suitable for use in all establishments,
including domestic establishments and those directly connectedto the public low-voltage power supply network that supplies
buildings used for domestic purposes.

31Appendix
Electromagnetic immunity
The RadiForce series has been tested at the following compliance levels (C) according to
the testing requirements (T) for professional healthcare facility environments specified in
IEC60601-1-2.
The customer or the user of the RadiForce series should ensure that it is used in such anenvironment.
Immunity test

Test level (T)Compliance level(C)

Electromagnetic environment GuidanceElectrostatic
discharge (ESD)
IEC61000-4-2±8 kV contact
discharge±8 kV contact
discharge

Floors should be wood, concrete
or ceramic tile. If floors are±15 kV air discharge ±15 kV air discharge covered with synthetic material,
the relative humidity should be atleast 30 %.

Electrical fasttransients / bursts
IEC61000-4-4±2 kV power lines±2 kV power lines±1 kV input / output
lines±1 kV input / output
lines

Surges
IEC61000-4-5±1 kV line to line

±1 kV line to line

±2 kV line to ground±2 kV line to ground

Voltage dips, shortinterruptions andvoltage variations on
power supply input
lines
IEC61000-4-11

0 % UT (100 % dip in 0 % UT (100 % dip in Mains power quality should beUT) 0.5 cycles and 1 UT) 0.5 cycles and 1 that of a typical commercial or
cycle
cycle
hospital environment. If the user
70 % UT (30 % dip in 70 % UT (30 % dip in of the RadiForce series requires
UT) 25 cycles / 50 Hz UT) 25 cycles / 50 Hz continued operation during power
mains interruptions, it is
0 % UT (100 % dip in 0 % UT (100 % dip in recommended that theUT) 250 cycles / 50
UT) 250 cycles / 50
RadiForce series be poweredHz
Hz
from an uninterruptible power
supply or a battery.

Power frequency
magnetic fields
IEC61000-4-8

30 A/m(50 / 60 Hz)

30 A/mMains power quality should bethat of a typical commercial or
hospital environment.
Mains power quality should bethat of a typical commercial or
hospital environment.

Power frequency magnetic fields
should be at levels characteristicof a typical location in a typicalcommercial or hospitalenvironment.
The product should be kept at
least 15 cm away from thesource of power frequency
magnetic fields during use.

32

Appendix

Immunity test

Test level (T)Compliance level(C)

Electromagnetic environment GuidancePortable and mobile RFcommunications equipment
should be used no closer to any
part of the RadiForce series,
including cables, than therecommended separation
distance calculated from theequation applicable to thefrequency of the transmitter.

Conducted
3 Vrmsdisturbances induced 150 kHz to 80 MHz
by RF fields
6 Vrms
IEC61000-4-6
ISM bands between
150 kHz and 80
MHz*1
Radiated RF fields
3 V/m
IEC61000-4-3
80 MHz – 2.7 GHz

3 VrmsRecommended separation
distanced = 1.2√P

6 Vrmsd = 1.2√P

3 V/m

d = 1.2√P, 80 MHz – 800 MHz
d = 2.3√P, 800 MHz – 2.7 GHz
Where "P" is the maximum
output power rating of thetransmitter in watts (W)according to the transmitter
manufacturer and "d" is therecommended separation
distance in meters (m). Field
strengths from fixed RF
transmitters, as determined by anelectromagnetic site survey*2,
should be less than thecompliance level in eachfrequency range*3.
Interference may occur in thevicinity of equipment marked withthe following symbol.

Note• UT is the a.c. mains voltage prior to application of the test level.
• At 80 MHz and 800 MHz, the higher frequency range applies.
• These guidelines regarding conducted disturbances induced by RF fields or radiated RF fields
may not apply in all situations. Electromagnetic propagation is affected by absorption andreflection from structures, objects and people.
*1 The ISM (industrial, scientific and medical) bands between 150 kHz and 80 MHz are 6.765 MHz to

6.795 MHz, 13.553 MHz to 13.567 MHz, 26.957 MHz to 27.283 MHz, and 40.66 MHz to 40.70
MHz.
*2 Field strengths from fixed transmitters, such as base stations for radio (cellular/cordless)telephones and land mobile radios, amateur radio, AM and FM radio broadcast and TV broadcast
cannot be predicted theoretically with accuracy. To assess the electromagnetic environment due
to fixed RF transmitters, an electromagnetic site survey should be considered. If the measured
field strength in the location in which the RadiForce series is used exceeds the applicable RFcompliance level above, the RadiForce series should be observed to verify normal operation. If

33

Appendix
abnormal performance is observed, additional measures may be necessary, such as reorienting or
relocating the RadiForce series.
*3 Over the frequency range 150 kHz to 80 MHz, field strengths should be less than 3 V/m.

34

Appendix
Recommended separation distances between portable or mobile RF communicationequipment and the RadiForce series
The RadiForce series is intended for use in an electromagnetic environment in whichradiated RF disturbances are controlled. The customer or the user of the RadiForce series
can suppress electromagnetic interference by maintaining a minimum distance (30 cm)
between portable and mobile RF communications equipment (transmitters) and the
RadiForce series. The RadiForce series has been tested at the following compliance level(C) for the requirements of the proximity electromagnetic field immunity test level (T) for thefollowing RF communication services.
Test
Bandwidth*1
frequency
(MHz)
(MHz)

Service*1

Modulation*2

Test level (T)*3(V/m)

Compliancelevel (C)
(V/m)

385

380 – 390TETRA 400

Pulse modulation*2 27
18 Hz

27

450

430 – 470

GMRS 460,
FRS 460

FM±5 kHz deviation
1 kHz sine

28

28

710

704 – 787

LTE Band 13, 17

Pulse modulation*2 9217 Hz9

800 – 960

GSM 800 / 900,TETRA 800,
iDEN 820CDMA 850,
LTE Band 5

Pulse modulation*2 28
18 Hz

28

1700 – 1990 GSM 1800;
CDMA 1900;
GSM 1900;
DECT;
LTE Band 1, 3, 4,
25;
UMTS

Pulse modulation*2 28
217 Hz28

2450

2400 – 2570 Bluetooth®,
WLAN,
802.11 b/g/n,
RFID 2450,
LTE Band 7

Pulse modulation*2 28
217 Hz28

5240

5100 – 5800 WLAN 802.11 a/n

Pulse modulation*2 9217 Hz

9

745
780
810
870
930
1720
1845
1970

5500
5785

*1 For some services, only the uplink frequencies are included.
*2 The carrier is modulated using a 50 % duty cycle square wave signal.
*3 Test levels were calculated with maximum power and 30 cm of separation distance.

35Appendix
The customer or the user of the RadiForce series suppress interference caused by proximity
magnetic fields by maintaining a minimum distance (15 cm) between RF transmitters andthe RadiForce series. The RadiForce series has been tested at the following compliancelevels (C) for the requirements of the proximity magnetic field immunity test levels (T).
Test frequency
134.2 kHz

Modulation
Pulse modulation*1

Test level (T)(A/m)

Compliance level (C)
(A/m)

65

65

7.5

7.5

2.1 kHz13.56 MHz

Pulse modulation*1
50 kHz*1 The carrier is modulated using a 50 % duty cycle square wave signal.

For other portable and mobile RF communication equipments (transmitters), minimum
distance between portable and mobile RF communications equipment (transmitters) and the
RadiForce series as recommended below, according to the maximum output power of thecommunications equipment.
Rated maximum
output power oftransmitter(W)

Separation distance according to frequency of transmitter(m)
150 kHz – 80 MHz
d = 1.2√P

80 MHz – 800 MHz
d = 1.2√P

800 MHz – 2.7 GHz
d = 2.3√P

0.01

0.12

0.12

0.23

0.1

0.38

0.38

0.73

1

1.2

1.2

2.3

10

3.8

3.8

7.3

100

12

12

23

For transmitters rated at a maximum output power not listed above, the recommended
separation distance "d" in meters (m) can be estimated using the equation applicable to thefrequency of the transmitter, where "P" is the maximum output power rating of thetransmitter in watts (W) according to the transmitter manufacturer.
Note• At 80 MHz and 800 MHz, the separation distance for a higher frequency range must be applied.
• These guidelines regarding conducted disturbances induced by RF fields or radiated RF fields
may not apply in all situations. Electromagnetic propagation is affected by absorption andreflection from structures, objects and people.

36Appendix

Information for Radio Interference
For U.S.A, Canada Only
FCC Supplier’s Declaration of Conformity
We, the Responsible Party
Company: EIZO Inc.
Address: 5710 Warland Drive, Cypress, CA 90630
Phone: (562) 431-5011
declare that the product
Trade name: EIZO
Model: RadiForce MX217
is in conformity with Part 15 of the FCC Rules. Operation of this product is subject to thefollowing two conditions: (1) this device may not cause harmful interference, and (2) thisdevice must accept any interference received, including interference that may cause
undesired operation.
This equipment has been tested and found to comply with the limits for a Class B digital
device, pursuant to Part 15 of the FCC Rules. These limits are designed to provide
reasonable protection against harmful interference in a residential installation. Thisequipment generates, uses, and can radiate radio frequency energy and, if not installed andused in accordance with the instructions, may cause harmful interference to radio
communications. However, there is no guarantee that interference will not occur in aparticular installation. If this equipment does cause harmful interference to radio or television
reception, which can be determined by turning the equipment off and on, the user is
encouraged to try to correct the interference by one or more of the following measures.
• Reorient or relocate the receiving antenna.
• Increase the separation between the equipment and receiver.
• Connect the equipment into an outlet on a circuit different from that to which the receiveris connected.
• Consult the dealer or an experienced radio/TV technician for help.
Changes or modifications not expressly approved by the party responsible for compliancecould void the user’s authority to operate the equipment.
Note
Use the attached specified cable below or EIZO signal cable with this monitor so as to keepinterference within the limits of a Class B digital device.
• AC Cord• Shielded Signal Cable (enclosed)
Canadian NoticeThis Class B information technology equipment complies with Canadian ICES-003.
Cet équipement informatique de classe B est conforme à la norme NMB-003 du Canada.
37

153 Shimokashiwano, Hakusan, Ishikawa 924-8566 Japan

Carl-Benz-Straße 3, 76761 Rülzheim, Germany

中国苏州市苏州工业园区展业路8号中新科技工业坊5B

1 Queens Square, Ascot Business Park, Lyndhurst Road,
Ascot, Berkshire, SL5 9FE, UK

Moosacherstrasse 6, Au, CH-8820 Wädenswil, Switzerland

www.eizoglobal.com
Copyright © 2022 - 2023 EIZO Corporation. All rights reserved.

00N0N343B1
IFU-MX217
2nd Edition – October 30th, 2023