21.3" Color LCD Monitor

A 2 megapixel medical monitor for display of CT/MRI
monochrome and 3D rendering/nuclear medicine color images.
Hybrid Display of Monochrome and Color

Boost Images for Easy Viewing

The Hybrid Gamma PXL function automaticallydistinguishes between monochrome and color
Hybridimages pixel by pixel, creating a hybrid display
Gamma PXL
where each pixel has optimum grayscale. As
a result, monochrome images such as CT, MRI
and X-ray are displayed in the ideal grayscale that corresponds
to DICOM® Part 14, while color images such as 3D rendering,nuclear medicine, ultrasound and endoscopy are faithfully
reproduced corresponding to Gamma 2.2. This helps improve
efficiency of viewing both monochrome and color images by
displaying them together on the one screen.

The Instant Backlight Booster function temporarily maxes thebrightness of the monitor for quickly viewing details in medicalimages. A single hotkey allows users to turn the function on for
multiple monitors at once so they can easily view more than one
screen under the same high-brightness conditions.
DICOM Part 14 is not supported while Instant Backlight Booster is on.

Automatically Distinguish & Display as Color
Gamma 2.2

Automatically Distinguish & Display as Monochrome
DICOM Part 14
Instant Backlight BoosterOﬀ

Instant Backlight BoosterOn

Specifications

Make the Precise Diagnosis
EIZO carefully measures and sets the grayscale at the factory
to ensure each monitor is compliant with DICOM Part 14.
Furthermore, at startup or upon wakeup, the EIZO patented drift
correction function quickly stabilizes the brightness level andcompensates the brightness fluctuations caused by the ambient
temperature and the passage of time, allowing medical images tobe faithfully reproduced with stable brightness and grayscale.

Achieve Clarity True to the Source Data
A medical monitor needs to be capable of high brightness inorder to meet performance standards. However, in order toachieve high brightness in an LCD panel, the pixel aperture
ratio has to be increased. This causes a typically unavoidable
decline in sharpness. With EIZO’s unique Sharpness Recovery
technology, the decrease in sharpness (MTF) is restored. This
allows you to display an image safely on the monitor that is true
to the original source data, even at high brightness levels.

Screen Variations for Specific User Needs
EIZO offers anti-glare (AG) and anti-reflection (AR) screen
variations to suit user environments. AG treatment is ideal for
exceptionally bright environments and drastically reduces glarefrom ambient lighting. AR treatment is ideal for moderately-lit
environments to reduce mild screen glare while maintaining
crisp text and images.

Maintain Image Quality Over TimeWith the Integrated Front Sensor (IFS) built into the front bezelof the RadiForce monitor and RadiCS LE software (included), you
can easily calibrate to DICOM Part 14 without having to mount,run, and remove an external sensor.
Simple calibration using the monitor backlight sensor is also supported.

Hassle-Free Multi-Monitor Configuration
Using the DisplayPort connection, you can drive several monitors
in a daisy chain sequence. This allows you to configure a multimonitor setup without the complicated hassle of excessive cabling.

Model Variations

RX270-BK: Anti-Glare coating, with stand, black
RX270-ARBK: Anti-Reflection coating, with stand,black

PanelColor (IPS)
LED21.3” (54.0 cm)
1200 x 1600 (3:4 aspect ratio)
324.0 x 432.0 mmTypeBacklight
Size
Native Resolution
Viewable Image Size(H x V)
Pixel Pitch (H x V)
Display Colors

0.270 x 0.270 mm
10-bit (DisplayPort): 1.07 billion froma palette of 543 billion (13-bit) colors
8-bit: 16.77 million from a palette of
543 billion (13-bit) colors
Viewing Angles (H / V, typical) 178° / 178°
Brightness (typical)
1000 cd/m²
Recommended
500 cd/m²
Brightness for Calibration
Contrast Ratio (typical) 1800:1
Response Time (typical) 20 ms (black-white-black)Video
Input Terminals
DisplayPort x 2, DVI-D
Signals
Output Terminals
DisplayPort (daisy chain)
Digital Scanning Frequency (H / V) 31 - 100 kHz / 59 - 61 HzUSBUpstream
USB 2.0: Type-B x 2Downstream
USB 2.0: Type-A x 2
Dedicated Charging Port USB Type-C ® (Power Supply 15 Wmax.)
Power
Power Input
AC 100 - 240 V: 50 / 60 Hz
Typical Power Consumption 33 W
Maximum Power Consumption 98 W
Power Save Mode1 W or lessSensorBacklight Sensor, Integrated Front
Sensor, Ambient Light Sensor
Features &Brightness Stabilization Yes
Functions
Digital Uniformity Equalizer Yes
Hybrid Gamma PXL
YesWork-and-Flow
Switch-and-Go, Point-and-Focus,
Instant Backlight Booster
Preset Modes
CAL Switch (DICOM, CAL1, CAL2,Custom, sRGB, Text)
OSD Languages
English, German, French, Italian,Japanese, Simplified Chinese, Spanish,Swedish, Traditional Chinese
PhysicalNet Weight
7.7 kgSpecifications Net Weight (Without Stand) 4.9 kg
Hole Spacing (VESA Standard) 100 x 100 mm
Certifications & Standards
CB, CE / UKCA (Medical Device), ANSI/(Please contact EIZO for the latest information)
AAMI ES60601-1, CAN/CSA-C22.2
No. 60601-1, IEC/EN60601-1, VCCI-B,
FCC-B, CAN ICES-3 (B), RCM, RoHS,
China RoHS, WEEE, CCC
FDA
510(k) Clearance for General
Radiography*DedicatedMonitor Quality
Supported
SoftwareControl Software
RadiCS
Supplied
Signal Cables
DisplayPort (3 m) x 2Accessories
OthersAC power cord (3 m), USB Type-A (May vary by country.
USB Type-B cable (3 m) x 2, Utility Disk
Please contact EIZO for(RadiCS LE, PDF instructions for use,details.)
PDF installation manual), instructions
for useWarranty
5 Years* Display of mammography images for diagnosis is not supported.

A graphics board that supports daisy chain is necessary.

Dimensions (Unit: mm)

Eye Relief with Gentle Light

5° 30°

464.5
482.3 - 572.3

70°
281

153 Shimokashiwano, Hakusan, Ishikawa 924-8566 Japan
Phone +81-76-277-6794, Fax +81-76-277-6793

https://www.eizoglobal.com

78°
90

RadiLight is an optional light that attaches to the back of aRadiForce monitor and illuminates the wall behind it. This
reduces eye strain for the radiologist viewing a bright monitor ina dark environment, while ensuring there are no reflections on
the screen to interfere with reading. It can be attached directly
to the monitor without removing the stand and does not take up
additional desk space.

356.5200

EIZO, the EIZO Logo, RadiForce and RadiCS are registered trademarks of EIZO Corporation in Japan and othercountries. RadiLight is trademark of EIZO Corporation. DICOM is the registered trademark of the National
Electrical Manufacturers Association for its standards publications relating to digital communications of medicalinformation. VESA is a registered trademark of Video Electronics Standards Association. USB Type-C is a registeredtrademark of USB Implementers Forum, Inc. All other company names, product names, and logos are trademarks
or registered trademarks of their respective owners. Specifications are subject to change without notice.
Copyright © 2023 EIZO Corporation. All rights reserved. (221004B)