Specifications

30.0" Color LCD Monitor

Model Variations

Achieve Clarity True to the Source Data
A medical monitor needs to be capable of high brightness inorder to meet performance standards. However, in order toachieve high brightness in an LCD panel, the pixel aperture
ratio has to be increased. This causes a typically unavoidable
decline in sharpness. With EIZO’s unique Sharpness Recovery
technology, the decrease in sharpness (MTF) is restored. This
allows you to display an image safely on the monitor that is true
to the original source data, even at high brightness levels.

Boost Images for Easy Viewing
With use of RadiCS LE software (included), the Instant Backlight
Booster function temporarily maxes the brightness of themonitor at the push of a button, allowing users to efficiently view
medical images which require higher brightness to distinguishdetails.
DICOM Part 14 is not supported while Instant Backlight Booster is on.

Instant Backlight Booster
Oﬀ

Instant Backlight Booster
OnConserve to Preserve
The RX670 is made of more than 77% recycled plastic. This
mitigates on the amount of plastic waste going into theenvironment, conserves resources, and promotes reuse ofmaterials for preserving the natural ecosystems.

RX670-BK: Anti-Glare coating, with stand, black
RX670-ARBK: Anti-Reflection coating, withstand, black
Panel
TypeColor (IPS)
Backlight
LEDSize30.0” (76.2 cm)
Native Resolution
3280 x 2048 (16:10 aspect ratio)
Viewable Image Size (H x V) 645.5 x 403.0 mm
Pixel Pitch (H x V)
0.1968 x 0.1968 mm
Display Colors
10-bit (DisplayPort TM): 1.07 billion
from a palette of 543 billion (13-bit)colors8-bit: 16.77 million from a palette of543 billion (13-bit) colors
Viewing Angles (H / V, typical) 178° / 178°
Brightness (typical)
1100 cd/m²
Recommended Brightness 500 cd/m²
for Calibration
Contrast Ratio (typical) 1800:1
Response Time (typical) 25 ms (black-white-black)Video
Input Terminals
USB Type-C (DisplayPort Alt Mode),
Signals
DisplayPort x 2, HDMI®
Output Terminals
USB Type-C (daisy chain)
Digital Scanning Frequency 31 - 127 kHz / 59 - 61 Hz(H / V)
USB
UpstreamUSB 2.0: Type-B x 2,
USB 2.0: Type-C (DisplayPort Alt Mode,
Power Delivery Source 94 W max.)
Downstream
USB 2.0: Type-A x 3
USB 2.0: Type-C (DisplayPort Alt Mode,
Power Delivery Source 15 W max.)
USB LAN Adapter
RJ-45 (1000BASE-T)
PowerPower Input
AC 100 - 240 V: 50 / 60 Hz
Power Consumption Typical / 61 W / 279 W / 0.5 W or lessMaximum / Power Save Mode
Backlight Sensor, Integrated Front Sensor,Sensor
Ambient Light Sensor
Features &Brightness Stabilization Yes
Functions
Digital Uniformity Equalizer Yes
Hybrid Gamma PXL
YesWork-and-Flow
Hide-and-Seek, Switch-and-Go,
Point-and-Focus, Instant Backlight Booster
Preset Modes
CAL Switch (DICOM, CAL, Patho, Custom,sRGB, Text)OSD Languages
English, German, French, Italian,Japanese, Simplified Chinese, Spanish,Swedish, Traditional Chinese
PhysicalNet Weight
15.8 kg / 11.5 kg (Without Stand)Specifications Hole Spacing (VESA Standard) 100 x 100 mm
Certifications & Standards
CE / UKCA (Medical Device), ANSI/(Please contact EIZO for the latest information)
AAMI ES60601-1, CAN/CSA-C22.2
No. 60601-1, IEC/EN60601-1, VCCI-B,
FCC-B, CAN ICES-3 (B), RCM, RoHS,
China RoHS, WEEE, CCC
510(k) Clearance for General Radiography*FDA
Dedicated
Monitor Quality Control Supported
Software RadiCS
SoftwareSignal Cables

SuppliedAccessories(May vary by country.
Please contact EIZO fordetails.)

USB Type-C (1.5 m), DisplayPort (3 m) x 2,HDMI (3 m)
AC power cord (3 m), USB Type-A - USB
Type-B cable (3 m) x 2, cable cover,Utility Disk (RadiCS LE, PDF installation
manual), instructions for use
5 Years

OthersWarranty* Display of mammography images for diagnosis is not supported.

The monitor uses molded pulp made from recycled cardboard
and newspaper in its packaging instead of conventional recycled
styrofoam or plastic, and cables are wrapped in paper instead ofplastic bags.

Dimensions (Unit: mm)
5°
682

30°
88441
490.5 - 590.5

Cushioning Environmental Impact

70°308.1

153 Shimokashiwano, Hakusan, Ishikawa 924-8566 Japan
Phone +81-76-277-6794

https://www.eizoglobal.com

225

EIZO, the EIZO Logo, RadiForce and RadiCS are registered trademarks of EIZO Corporation in Japan and other countries.
The terms HDMI, HDMI High-Definition Multimedia Interface, HDMI trade dress and the HDMI Logos are trademarks orregistered trademarks of HDMI Licensing Administrator, Inc. DisplayPort, the DisplayPort Compliance Logo and VESAare trademarks of the Video Electronics Standards Association in the United States and other countries. USB Type-C isa registered trademark of USB Implementers Forum, Inc. DICOM is the registered trademark of the National Electrical
Manufacturers Association for its standards publications relating to digital communications of medical information. Allother company names, product names, and logos are trademarks or registered trademarks of their respective owners.
Specifications are subject to change without notice.
Copyright © 2024 EIZO Corporation. All rights reserved. (240801)

6 Megapixel High-Brightness Multi-Modality Monitor
with USB Type-C Docking and Comfort Light for Medical Image Reading
Multi-Modality Capability with 6MP Resolution
The spacious 30-inch screen displays 6 megapixels (3280 x 2048pixels) of information. Multi-modality capability enables thesimultaneous display of examination images, such as chest X-ray,CT, endoscopy, or ultrasound, offering a high degree of freedom
in the layout and facilitating efficient image reading.

Hybrid Display of Monochrome and Color

Streamlined Connection with USB Type-C

Hassle-Free Multi-Monitor Setup

The Hybrid Gamma PXL function automatically distinguishes
between monochrome and color images pixel by pixel, creating a
hybrid display where each pixel has optimum grayscale.

The monitor features a USB Type-C® connectivity that allows youdisplay video, supply power to connected mobile workstations,stable wired network connection, and USB device connection.

The monitor’s USB Type-C connection supports daisy chaining
for easily connecting multiple monitors from a single USB Type-C
video signal output of mobile workstations. This allows you toconfigure a multi-monitor setup without hassle of excess cabling.

6MP 3280 x 2048 Pixels
1536 Pixels

3MP

2048 Pixels

2048 Pixels

3MP

1536 Pixels

Automatically Distinguish & Display as Color
Gamma 2.2

Automatically Distinguish & Display as Monochrome
DICOM Part 14

Resolution Comparison

Make the Precise Diagnosis

Maintain Image Quality Over Time

Eye Relief with Comfort Light

Easily Navigate Your Workspace

EIZO carefully measures and sets each grayscale tone to createa monitor compliant with DICOM® Part 14. This ensures the mostconsistent shading possible, allowing for the most accurate
diagnosis.

With the Integrated Front Sensor (IFS) built into the front bezelof the RadiForce monitor and RadiCS LE software (included), you
can easily calibrate to DICOM Part 14 without having to mount,run, and remove an external sensor.

The built-in comfort light gently illuminates the wall behind themonitor without interfering with the low ambient lighting typicalof reading rooms. This reduces the contrast between the walland monitor to prevent eye fatigue without creating glare on thescreen.

The monitor is also equipped with a spotlight which allows you
to see printed documents or your keyboard. The position can beeasily adjusted in order to achieve the ideal lighting.

Simple calibration using the monitor backlight sensor is also supported.

Without Integrated Calibration Sensor

With Integrated Front Sensor (IFS)Specifications

30.0" Color LCD Monitor

Model Variations

Achieve Clarity True to the Source Data
A medical monitor needs to be capable of high brightness inorder to meet performance standards. However, in order toachieve high brightness in an LCD panel, the pixel aperture
ratio has to be increased. This causes a typically unavoidable
decline in sharpness. With EIZO’s unique Sharpness Recovery
technology, the decrease in sharpness (MTF) is restored. This
allows you to display an image safely on the monitor that is true
to the original source data, even at high brightness levels.

Boost Images for Easy Viewing
With use of RadiCS LE software (included), the Instant Backlight
Booster function temporarily maxes the brightness of themonitor at the push of a button, allowing users to efficiently view
medical images which require higher brightness to distinguishdetails.
DICOM Part 14 is not supported while Instant Backlight Booster is on.

Instant Backlight Booster
Oﬀ

Instant Backlight Booster
OnConserve to Preserve
The RX670 is made of more than 77% recycled plastic. This
mitigates on the amount of plastic waste going into theenvironment, conserves resources, and promotes reuse ofmaterials for preserving the natural ecosystems.

RX670-BK: Anti-Glare coating, with stand, black
RX670-ARBK: Anti-Reflection coating, withstand, black
Panel
Type
Color (IPS)
Backlight
LEDSize30.0” (76.2 cm)
Native Resolution
3280 x 2048 (16:10 aspect ratio)
Viewable Image Size (H x V) 645.5 x 403.0 mm
Pixel Pitch (H x V)
0.1968 x 0.1968 mm
Display Colors
10-bit (DisplayPort TM): 1.07 billion
from a palette of 543 billion (13-bit)colors8-bit: 16.77 million from a palette of543 billion (13-bit) colors
Viewing Angles (H / V, typical) 178° / 178°
Brightness (typical)
1100 cd/m²
Recommended Brightness 500 cd/m²
for Calibration
Contrast Ratio (typical) 1800:1
Response Time (typical) 25 ms (black-white-black)Video
Input Terminals
USB Type-C (DisplayPort Alt Mode),
Signals
DisplayPort x 2, HDMI®
Output Terminals
USB Type-C (daisy chain)
Digital Scanning Frequency 31 - 127 kHz / 59 - 61 Hz(H / V)
USBUpstream
USB 2.0: Type-B x 2,
USB 2.0: Type-C (DisplayPort Alt Mode,
Power Delivery Source 94 W max.)
Downstream
USB 2.0: Type-A x 3
USB 2.0: Type-C (DisplayPort Alt Mode,
Power Delivery Source 15 W max.)
USB LAN Adapter
RJ-45 (1000BASE-T)
Power
Power Input
AC 100 - 240 V: 50 / 60 Hz
Power Consumption Typical / 61 W / 279 W / 0.5 W or lessMaximum / Power Save Mode
Backlight Sensor, Integrated Front Sensor,
Sensor
Ambient Light Sensor
Features &Brightness Stabilization Yes
Functions
Digital Uniformity Equalizer Yes
Hybrid Gamma PXL
YesWork-and-Flow
Hide-and-Seek, Switch-and-Go,Point-and-Focus, Instant Backlight Booster
Preset Modes
CAL Switch (DICOM, CAL, Patho, Custom,sRGB, Text)
OSD Languages
English, German, French, Italian,Japanese, Simplified Chinese, Spanish,Swedish, Traditional Chinese
PhysicalNet Weight
15.8 kg / 11.5 kg (Without Stand)Specifications Hole Spacing (VESA Standard) 100 x 100 mm
Certifications & Standards
CE / UKCA (Medical Device), ANSI/(Please contact EIZO for the latest information)
AAMI ES60601-1, CAN/CSA-C22.2
No. 60601-1, IEC/EN60601-1, VCCI-B,FCC-B, CAN ICES-3 (B), RCM, RoHS,China RoHS, WEEE, CCC
510(k) Clearance for General Radiography*FDA
DedicatedMonitor Quality Control Supported
Software RadiCS
SoftwareSignal Cables

Supplied
Accessories(May vary by country.
Please contact EIZO fordetails.)

USB Type-C (1.5 m), DisplayPort (3 m) x 2,HDMI (3 m)
AC power cord (3 m), USB Type-A - USB
Type-B cable (3 m) x 2, cable cover,Utility Disk (RadiCS LE, PDF installation
manual), instructions for use
5 Years

OthersWarranty* Display of mammography images for diagnosis is not supported.

The monitor uses molded pulp made from recycled cardboardand newspaper in its packaging instead of conventional recycled
styrofoam or plastic, and cables are wrapped in paper instead ofplastic bags.

Dimensions (Unit: mm)
5°
682

30°
88441
490.5 - 590.5

Cushioning Environmental Impact

70°
308.1

153 Shimokashiwano, Hakusan, Ishikawa 924-8566 Japan
Phone +81-76-277-6794

https://www.eizoglobal.com

225

EIZO, the EIZO Logo, RadiForce and RadiCS are registered trademarks of EIZO Corporation in Japan and other countries.
The terms HDMI, HDMI High-Definition Multimedia Interface, HDMI trade dress and the HDMI Logos are trademarks orregistered trademarks of HDMI Licensing Administrator, Inc. DisplayPort, the DisplayPort Compliance Logo and VESA
are trademarks of the Video Electronics Standards Association in the United States and other countries. USB Type-C isa registered trademark of USB Implementers Forum, Inc. DICOM is the registered trademark of the National Electrical
Manufacturers Association for its standards publications relating to digital communications of medical information. Allother company names, product names, and logos are trademarks or registered trademarks of their respective owners.
Specifications are subject to change without notice.
Copyright © 2024 EIZO Corporation. All rights reserved. (240801)