Every life is unique. Every person‘s medical treatment should be tailored to meettheir individual needs.
In the age of precision medicine, the possibilities offered by biotechnologies,
artificial intelligence, and information technology open up completely new avenues
for diagnosis, prevention, and treatment.
Precision requires comprehensive information. Collecting, linking, and analyzing
data, as well as recording, storing, and evaluating image data therefore
represents a critical resource for modern medical practices.
Faster treatment success, better quality of life: Technical innovation has an
immediate impact on the medical processes in hospitals and operating rooms.
Which is why we employ all of our experience and work together with highly
qualified medical teams to produce reliable systems for processing image datain the age of precision medicine.
Our knowledge is in the service of better health.
Every life is worth it.
Making Each Life Visual.

2

3

RadiForce specially designed 1 to 12 megapixel
monochrome and color monitors take full accountof medical institutions' need for different typesof monitors with DICOM® Part 14 standard
calibration and high-performance capabilities
required for precise diagnoses.

View at the Appropriate Resolution

Make the Precise Diagnosis

Manage Effortless Quality Control

Each modality varies in its display of medical images
with regards to size and information volume.
RadiForce monitors come in a range of resolutions
for displaying images appropriate for each modality.

EIZO carefully measures and sets
each grayscale tone for compliancewith DICOM Part 14. Furthermore,
at startup or upon wakeup, the drift
correction function quickly stabilizes
the brightness level and compensates the brightness
fluctuations caused by the ambient temperature and thepassage of time, allowing medical images to be faithfully
reproduced with stable brightness and grayscale.

An Integrated Front Sensor (IFS) housed within the front
bezel measures brightness and grayscale tones andcalibrates to the DICOM Part 14 standard. The hands-free
IFS performs quality control tasks and does not interferewith the viewing area while in use. This dramaticallycuts the workload and maintenance costs needed formaintaining monitor quality control.

2150

Original Data

1000

600

512

512

CT

1000

800

US

2294

1914

1760

Angiography

Mammography

Chest CR

All models except the MX243W, MX194, and MS236WT-A.

MS236WT-A features a DICOM preset mode for optimal
medical image viewing.

Monitor Resolution
4200
2048

4096

1600

2800

2160

2560

2048

1200

2048

3280

1536

Without IFS

With IFS

4

5

Uniformity Across the Screen

Select the Ideal Mode for Modalities

Variations for Specific User Needs

Display Both Monochrome and ColorThe Digital Uniformity Equalizer (DUE) function
helps to even out fluctuations in brightness and chroma
on different parts of the screen to provide smootherimages, a quality typically difficult to attain due tothe characteristics of LCD monitors.

The CAL Switch function allows you to choose various
modes for different modalities such as CR, CT, andendoscopy. It can be conveniently accessed using themonitor's front panel buttons to easily switch to optimal
image viewing conditions.

All models except the MS236WT-A.

Number or type of the modes vary by model. Check the specifications
on pages 20 - 23.

EIZO offers anti-glare (AG) and anti-reflection (AR)screen variations to suit user environments. AG
treatment is ideal for exceptionally bright environments
and drastically reduces glare from ambient lighting.
AR treatment is ideal for moderately-lit environments
to reduce mild screen glare while maintaining crisp
text and images.

The Hybrid Gamma PXL function
automatically creates a hybrid display
Hybridwhere each pixel has optimum grayscale.
Gamma PXLAs a result, monochrome images suchas x-ray, MRI and CT are displayed in theideal DICOM Part 14 grayscale, while color images suchas ultrasound and endoscopy are reproduced
corresponding to Gamma 2.2. This improves theefficiency of viewing both monochrome and color
images together on the one screen.

CT ModeCR ModeIncident lightReﬂected lightLight from backlight

Without DUE

DICOM Part 14

AG

Incident light

Gamma 2.2

Reﬂected lightLight from backlight

Image for illustrative
purposes only.

6

Check the specifications on pages 20 - 23 for availability.

With DUE

US ModeES ModeAutomatically Distinguish DICOM Part 14& Display as Monochrome
Automatically Distinguish Gamma 2.2& Display as Color

AR

7

Conserve Energy While Away

Stay Confident with Stable Brightness

Improve Operability

Effortless Installation

The presence sensor equipped with some models
prompts the monitor to switch to power save mode
when it detects you are away, and then resumes
normal operation when you return. This ensures thatthe monitor conserves power when it is not in use,
uniting convenience with savings.

EIZO's confidence in its product quality extends tobrightness stability which is also covered during theusage time specified in the warranty.

EIZO's highly versatile stand offers tilt, swivel, anda wide height adjustment range, enabling you to use
the monitor with greater comfort.

All models except the MS236WT-A.

The specific moving ranges of stands varies by model. See pages
20 - 23 for specifications of all monitors.

EIZO, in collaboration with business partners, verifies
the compatibility of healthcare workstations and desktop
PCs with EIZO monitors. With our years of experienceand know-how, we undertake professional testing on
new workstations and PCs as soon as they are released.
In the healthcare field where reliability is everything,
EIZO is providing the assurance needed for effortless
installation.

Check the specifications on pages 20 - 23 for availability.

90 mm
90°

35°

70°

We verify aspects such as:
Stable operation with workstations/PCs
Image quality that can display DICOM medical images

Other Monitors

Height

2024

2025

2026

Tilt

Rotation

SwivelPictured: RX370

15°

RadiForce Monitors

8

MS236WT-A comes with astand that lets you tilt themonitor back for easy touch
pen use.

70°

9With advances in medical imaging
technology over the years, hospitals
are now handling a wider variety andlarger volume of image data. The multimodality approach of RadiForce superhigh-resolution diagnostic monitors
allows a variety of images to be displayedon a single screen — an essential step

6MP

forward for medicine.

30.0" Color LCD Monitor

Evolve Your Image Reading

12MP

30.9" Color LCD Monitor

As more image modalities become digitalized, radiologists are viewing
an increasing amount of information on their screens. EIZO's unique
Work-and-Flow technology alleviates the complexity of the imaging
workﬂow with new functions developed with the radiologist in mind.
Users can take advantage of Work-and-Flow features with the RadiForcemonitors and bundled RadiCS LE software.

See more with animations.
https://www.eizoglobal.com/i/workandflow/

Multi-Modality ReadinessSeamlessly View Images

Quick Referencing

Barrier-Free WorkstyleMulti-modality monitors are capable of displaying
images to suit a number of modalities such as CR,
DR, MRI, CT, and ultrasound. With multi-modality
support, you can increase work efficiency with theability to view numerous medical images on one
screen with exceptional accuracy.

RadiForce multi-modality monitors allow you to view
images side by side without the obtrusive bezels typicallyfound in a multi-monitor setup. This prevents the eyefrom being disrupted when moving between two screens
for reader efficiency.

The Hide-and-Seek function enables users to easily hide
the PinP (Picture in Picture) window not currently in use
and reopen it as needed by moving the mouse cursor
to the edge of the screen. This eliminates the need foran extra monitor while still allowing quick and efficient
viewing of reports, patient charts, and other information.

With the Switch-and-Go function, you can operate twodifferent workstations at the same time with a single
mouse and keyboard. Work across several monitors withintuitive cursor movement or switch signals between
workstations as needed without changing your mouse or
keyboard each time. This makes it possible to reduce thenumber of monitors in the workflow and improves work
efficiency.

Check the specifications on pages 20 - 23 for availability.

Check the specifications on pages 20 - 23 for availability.

BeforeWS

Patient
List

Image Viewer

After

10

Before
WS1

Patient
Chart

Image Viewer

Patient
List

Image Viewer

WS2

After

11

MammoDuo integrates two 5 megapixel monitorsside by side on a specifically designed stand.

With the world's narrowest bezel of 7.5 mm on a 5 megapixel monitor, two monitorsside by side have a combined bezel width of only 15 mm. Furthermore the bezel is only
2.5 mm thick to help your eyes swiftly move from one monitor to another.

It is vital in the process of early breast
cancer detection that monitors display
accurate and consistent quality images.
EIZO provides optimum diagnosis
confidence with distinctive versions of
the RadiForce Mammo-Series breast
imaging monitors for displaying breast
screening images.

5MP5MP

12MP

5MP

30.9" Color LCD Monitor

See more with animations.
https://www.eizoglobal.com/i/workandflow/

Quick and Easy Focus
With the Point-and-Focus function, you can quickly
select and focus areas of concern with just yourmouse and keyboard. Change the brightness andgrayscale tones of certain points on the screen to
make interpretation easier.

21.3" Color LCD Monitors with Dual Screen Conﬁguration

21.3" Color LCD Monitor

5MP5MP

21.3" Monochrome LCD Monitors with Dual Screen Conﬁguration

5MP

21.3" Monochrome LCD Monitor

All-in-One Breast Imaging

Full Color Support

Optimum Breast Screening

The RadiForce RX1270 creates the perfect balance
between comfort and functionality in reading rooms.
With its 12 megapixel (4200 x 2800) resolution andcompact 30.9-inch size, you can comfortably view
several breast images side by side on a single screen.
Furthermore, the monitor comes with a rear light whichgently illuminates the wall behind, creating the ideal
ambient lighting for improved reading accuracy.

As the world's first medical monitor with an LTPS (lowtemperature polysilicon) panel, the RX560 achieves amaximum brightness of 1100 cd/m2 and a contrast ratioof 1500:1 similar to that of monochrome monitors. This
ensures that with a single screen, monochrome images
such as breast tomosynthesis and mammography aredisplayed accurately alongside color images such as MRI,
CT, ultrasound, pathology, and biopsies to accurately
examine breast tissue.

The 5 megapixel (2048 x 2560) GX560 adopts an LTPS
(low temperature polysilicon) panel with a maximum
brightness of 2500 cd/m2 and a pixel pitch of 0.165 mm.
It reproduces large volume mammography images
accurately with minimal thinning and patchiness, andis suitable for distinguishing spiculated masses andthe delicate shadows of calcifications. Furthermore, 12
millisecond response time allows smooth and efficient
viewing of breast tomosynthesis.

Check the specifications on pages 20 - 23 for availability.
BeforeAfter

Breast Tomosynthesis

12

Mammography

13

High-resolution 3 megapixel
monitors are capable offully displaying chest X-ray
images. 2 megapixel
monitors are ideal for a wide
variety of tasks from viewing
CR, DR, MRI, and CT images
to use as a PACS / HIS / RIS
terminal.

3MP

See more with animations.
https://www.eizoglobal.com/i/workandflow/

Boost Images for Easy Viewing
The Instant Backlight Booster function temporarily
maxes the brightness of the monitor for quickly
making detailed medical images easier to see.

2MP

21.3" Color LCD Monitor

21.3" Color LCD Monitor

3MP

2MP

21.3" Monochrome LCD Monitor

21.3" Monochrome LCD Monitor

Achieve Clarity True to the Source Data

Hassle-Free Multi-Monitor Configuration

Discern Subtleties in Grayscale Tones

A medical monitor needs to be capable of highbrightness in order to meet performance standards.
However, in order to achieve high brightness in an LCD
panel, the pixel aperture ratio has to be increased. This
causes an unavoidable decline in sharpness. With EIZO's
unique Sharpness Recovery technology, the decrease in
sharpness (MTF) is restored. This allows you to display
an image safely on the monitor that is true to the original
source data, even at high brightness levels.

Utilizing the DisplayPort™ output connection, you candrive several monitors in a daisy chain sequence. This
allows you to configure a multi-monitor setup without thecomplicated hassle of excessive cabling.

10-bit (1,024 tones) simultaneous grayscale display
reproduces monochrome images with a high bit-depth
for a sharper, clearer result.

Available with the RX1270, RX660, RX560, GX560, RX370, and RX270.

10-bit graphics board and 10-bit viewer software needed for
10-bit display.

In

Available with the RX1270, RX660, RX560, GX560, RX370, and RX270.
In

DICOM Part 14 is not supported while Instant Backlight
Booster is on.
Check the specifications on pages 20 - 23 for availability.

Original Data

In

Sharpness
Sharpness
Recovery Oﬀ Recovery On

Displayed
Image

Without DisplayPort Output

8-bit Displayed Image

White

10-bit Displayed Image

Black

Sharpness
improves
by approx.
50% or more

MTF

1.0

In

Sharpness Recovery On
Sharpness Recovery Oﬀ
0

14

Instant Backlight Booster
Oﬀ

Instant Backlight Booster
On

0.2

0.4

0.6

0.8

1

1.2

1.4

1.6

Out

In

Out

In

With DisplayPort Output

Spatial Frequency (cycles/mm)
RX370

15

Superior cost performance monitors are ideal for viewing patient charts with MRI and CT
medical images in DICOM Part 14 standard. In addition, they are available in widescreen andsquare formats in various resolutions to meet the diverse needs of hospitals and clinics.

1MP

2.3MP

8MP

30.5" Color LCD Monitor

2MP

19.0" Color LCD Monitor

24.1" Color LCD Monitor

21.3" Color LCD Monitor

2MP

23.0" Touch Panel Color LCD Monitor

Streamlined Connection with USB Type-C

Enhanced Workflow with 8MP Resolution

Rotate the Monitor According to the Image

Smooth and Detailed Handwriting

The MX317W features a USB Type-C® connectivity thatallows you display video, transmit USB signals, supply
power to connected mobile workstations, and eventransmit a network signal. This convenient multi-purpose
connectivity makes it faster and easier to connect yourmobile workstation when working remotely or at home.

The MX317W offers the highest resolution from the
MX-Series, displaying 8 megapixels of information (4096
x 2160 pixels) on the large 30.5-inch screen. In medical
settings where multiple applications and windows areoften displayed simultaneously, this allows for freelayout on a single screen.

After installing the included RadiCS LE quality control
software, you can link the Image Rotation Plus function
with the MX217's built-in gravity sensor. This enables
the screen to automatically switch to either portrait
or landscape mode based on the orientation of themonitor.

The MS236WT-A accepts touch input from a bare finger orcommercially-available stylus pen, so small and detailed
letters can easily be written into a medical record.

Resolution

FullHD (1920 x 1080)
WQHD (2560 x 1440)

Rotate from Portrait to Landscape

4K (4096 x 2160)

The MS236WT-A is equipped with palm rejection whichallows you to rest your hand directly on the screen
without causing any unintended touch input, so that you
can focus on your writing.
Palm rejection minimum activation area is 2 × 2 cm.

Rotate from Landscape to Portrait

16

17

Examination Room
Reading RoomWith filmless imaging spreading in medicine, maintaining the quality ofmonitors for medical imaging is becoming increasingly important. With theknow-how and experience as a specialist in visual display solutions, EIZO offers

Control Roommonitor quality control solutions for diagnostic precision and comprehensive
management to contribute to the improvement of the quality of medical care.

Reception
Consultation Room
Server Room

Client Software
Server Software

Monitor Quality Control Software& Calibration Sensor

Network QC
Management Software

EIZO

Maintain Quality Control
of Individual Monitors

Maintain Quality Control
for a Large Number of Monitors

Ensuring that the quality control of each client monitor
complies with important medical standards, fromcalibration to acceptance and constancy tests to history
and asset management, requires technical know-how
and experience. EIZO offers software and sensors thatmake quality control efficient and user-friendly.

Maintaining quality control of a large number of monitors
in hospitals calls for a lot of effort. EIZO offers centralized
management of client monitors connected to the hospital
network, providing increased efficiency of monitor QC
operations.
Hosting Service

Network QC Management Server Provider

Expert Quality Control
Services for Reassurance
Setting up and maintaining a server for monitor
quality control operations is a significant investment. EIZO will
setup and host the web server for you for efficient centralized
control of all connected monitors.

18

19

SPECIFICATIONS
12MP
RX1270-BK: Anti-Glare coating, with stand, black
RX1270-ARBK: Anti-Reflection coating, with stand, black

RX660-BK: Anti-Glare coating, with stand, black
RX660-ARBK: Anti-Reflection coating, with stand, black

RX560-BK-MD: Anti-Glare coating, two screens, with dual stand, black
RX560-ARBK-MD: Anti-Reflection coating, two screens, with dual stand, black
RX560-BK: Anti-Glare coating, one screen, with stand, black
RX560-ARBK: Anti-Reflection coating, one screen, with stand, black

GX560-BK-MD: Anti-Glare coating, two screens, with dual stand, black
GX560-ARBK-MD: Anti-Reflection coating, two screens, with dual stand, black
GX560-BK: Anti-Glare coating, one screen, with stand, black
GX560-ARBK: Anti-Reflection coating, one screen, with stand, black

Type

Color (IPS)Color (IPS)Color (IPS)

Monochrome (IPS)

Backlight

LED

LED

LED

LED

Size

30.9" (78.4 cm)

30.0" (76 cm)

21.3" (54.1 cm)

21.3" (54.1 cm)

Native Resolution

4200 x 2800 (3:2 aspect ratio)

3280 x 2048 (16:10 aspect ratio)

2048 x 2560 (4:5 aspect ratio)

2048 x 2560 (4:5 aspect ratio)

Viewable Image Size (H x V)

652.7 x 435.1 mm

645.5 x 403.0 mm

337.9 x 422.4 mm

337.9 x 422.4 mm

Pixel Pitch (H x V)

0.1554 x 0.1554 mm

0.1968 x 0.1968 mm

0.165 x 0.165 mm

0.165 x 0.165 mm

Display Colors / Grayscale Tones

10-bit (DisplayPort): 1.07 billion from a palette of 543 billion (13-bit) colors
8-bit: 16.77 million from a palette of 543 billion (13-bit) colors

10-bit (DisplayPort): 1.07 billion from a palette of 543 billion (13-bit) colors
8-bit: 16.77 million from a palette of 543 billion (13-bit) colors

10-bit (DisplayPort): 1.07 billion from a palette of 543 billion (13-bit) colors
8-bit: 16.77 million from a palette of 543 billion (13-bit) colors

10-bit (DisplayPort): 1,024 from a palette of 16,369 (14-bit) tones
8-bit: 256 from a palette of 16,369 (14-bit) tones

Viewing Angles (H / V, typical)

178° / 178°

176° / 176°

178° / 178°

178° / 178°

Brightness (typical)

1200 cd/m2

1000 cd/m²

1100 cd/m²

2500 cd/m²

Recommended Brightness for Calibration

500 cd/m2

500 cd/m²

500 cd/m²

1000 cd/m², 600 cd/m²

Contrast Ratio (typical)

1500:1

1500:1

1500:1

1700:1

Response Time (typical)

12 ms (black-white-black)25 ms (black-white-black)

12 ms (black-white-black)

12 ms (black-white-black)

Input Terminals

DisplayPort x 2, HDMI®

DisplayPort x 2, DVI-D (dual link)

DisplayPort, DVI-D (dual link)

DisplayPort x 2, DVI-D (dual link)

Output Terminals―

DisplayPort (daisy chain)

DisplayPort (daisy chain)

DisplayPort (daisy chain)

Digital Scanning Frequency (H / V)

31 - 175 kHz / 29 - 61 Hz

31 - 127 kHz / 22 - 61 Hz

31 - 135 kHz / 23 - 61 Hz

USB

31 - 135 kHz / 23 - 61 Hz

UpstreamUSB 2.0: Type-B x 2

USB 2.0: Type-B x 2

USB 2.0: Type-B

USB 2.0: Type-B x 2

Downstream

USB 2.0: Type-A x 3

USB 2.0: Type-A x 3

USB 2.0: Type-A x 2

USB 2.0: Type-A x 2

USB LAN Adapter

―

―

―

―

Dedicated Charging Port

―

―

―

Power

―

Power Input

AC 100 - 240 V: 50 / 60 Hz

AC 100 - 240 V: 50 / 60 Hz

AC 100 - 240 V: 50 / 60 Hz

AC 100 - 240 V: 50 / 60 Hz

Typical Power Consumption

77 W

93 W

43 W

28 W

Maximum Power Consumption

188 W

190 W

87 W

79 W

Power Save Mode2 W or less

1.6 W or less

1 W or lessSensorFeatures &Functions

1 W or less

Backlight Sensor, Integrated Front Sensor, Ambient Light Sensor

Backlight Sensor, Integrated Front Sensor, Presence Sensor, Ambient Light Sensor

Backlight Sensor, Integrated Front Sensor, Presence Sensor, Ambient Light Sensor

Backlight Sensor, Integrated Front Sensor, Presence Sensor, Ambient Light Sensor

Brightness Stabilization

Yes

Yes

Yes

Yes

Digital Uniformity Equalizer

Yes

Yes

Yes

Yes

Hybrid Gamma PXL

Yes

Yes

Yes

―

Work-and-Flow

Hide-and-Seek, Switch-and-Go, Point-and-Focus, Instant Backlight Booster

Hide-and-Seek, Switch-and-Go, Point-and-Focus

Point-and-Focus

Switch-and-Go, Point-and-Focus

Preset Modes

CAL Switch (DICOM, CAL1, CAL2, Custom, sRGB, Text)

CAL Switch (DICOM, CAL1, CAL2, Custom, sRGB, Text)

CAL Switch (DICOM, CAL1, CAL2, Custom, sRGB, Text)

CAL Switch (DICOM, CAL1, CAL2, Text)

OSD Languages

English, German, French, Italian, Japanese, Simplified Chinese, Spanish,
Swedish, Traditional ChineseEnglish, German, French, Italian, Japanese, Simplified Chinese, Spanish,
Swedish, Traditional ChineseEnglish, German, French, Italian, Japanese, Simplified Chinese, Spanish, Swedish,Traditional ChineseEnglish, German, French, Italian, Japanese, Simplified Chinese, Spanish, Swedish, Traditional
ChinesePhysicalSpecifications

Net Weight

15.6 kg

14.2 kg

RX560-MD, RX560-AR-MD: 17.3 kg
RX560, RX560-AR: 8.1 kg

GX560-MD, GX560-AR-MD: 17.1 kg
GX560, GX560-AR: 8 kg

Net Weight (Without Stand)11.5 kg

10.1 kg

5.3 kg

5.2 kgHole Spacing (VESA Standard)

100 x 100 mm

100 x 100 mm

100 x 100 mm

100 x 100 mm

CE / UKCA (Medical Device), ANSI/AAMI ES60601-1, CAN/CSA-C22.2 No. 606011, IEC/EN60601-1, VCCI-B, FCC-B, CAN ICES-3 (B), RCM, RoHS, China RoHS,
WEEE, CCC, EAC

CE / UKCA (Medical Device), ANSI/AAMI ES60601-1, CAN/CSA-C22.2 No. 606011, IEC/EN60601-1, VCCI-B, FCC-B, CAN ICES-3 (B), RCM, RoHS, China RoHS,
WEEE, CCC, EAC

RX560, RX560-AR:
CE / UKCA (Medical Device), ANSI/AAMI ES60601-1, CAN/CSA-C22.2 No. 60601-1,
IEC/EN60601-1, VCCI-B, FCC-B, CAN ICES-3 (B), RCM, RoHS, China RoHS, WEEE,
CCC, EAC

GX560, GX560-AR:
CE / UKCA (Medical Device), ANSI/AAMI ES60601-1, CAN/CSA-C22.2 No. 60601-1, IEC/
EN60601-1, VCCI-B, FCC-B, CAN ICES-3 (B), RCM, RoHS, China RoHS, WEEE, CCC, EAC

Certifications & Standards 1

510(k) Clearance for Breast Tomosynthesis, Mammography, and General
Radiography

510(k) Clearance for General Radiography

510(k) Clearance for Breast Tomosynthesis, Mammography, and General
Radiography

510(k) Clearance for Breast Tomosynthesis, Mammography, and General Radiography

Monitor Quality Control Software RadiCS

Supported

Supported

Supported

Supported

Signal Cables

DisplayPort (3 m) x 2, HDMI (2 m)Dual Link DVI-D (3 m), DisplayPort (3 m) x 2, DisplayPort (0.28 m)

RX560-MD, RX560-AR-MD: Dual Link DVI-D (3 m) x 2, DisplayPort (3 m) x 2,
DisplayPort (1 m)

GX560-MD, GX560-AR-MD: DisplayPort (3 m) x 4, DisplayPort (1 m)

OthersAC power cord (3 m), USB Type-A - USB Type-B cable (3 m) x 2, cable cover,Utility Disk (RadiCS LE, PDF installation manual), instructions for use

AC power cord (3 m), USB Type-A - USB Type-B cable (3 m) x 2, cable cover,Utility Disk (RadiCS LE, PDF installation manual), instructions for use

FDA 1, 2, 3
Dedicated
SoftwareSupplied
Accessories 4

5MP5MP

RX560, RX560-AR: Dual Link DVI-D (3 m), DisplayPort (3 m)

5 Years

1
2
3
4

20

15°
40°

RX1270, RX660,
RX560, GX560

RX560-MD
GX560-MD

308

30°
88

5 Years

5 Years

RX560-MD

5° 25°
709

225

308

78

RX560

5° 30 °
78

354.5

GX560-MD

5° 25°
709

78

GX560

90°

225

308

225

281

5° 30 °
78

354.5
90°

452
476 - 566

476
508 - 608

15°
70°

5°
682.5

GX560, GX560-AR:
AC power cord (3 m), USB Type-A - USB Type-B cable (3 m) x 2, Utility Disk (RadiCS LE, PDF
installation manual), instructions for use

441
490.5 - 590.5

5° 30°
88.6

689.8

RX560, RX560-AR:
AC power cord (3 m), USB Type-A - USB Type-B cable (3 m), Utility Disk (RadiCS LE,
PDF installation manual), instructions for use

452
476 - 566

5 Years

GX560-MD, GX560-AR-MD:
AC power cord (3 m) x 2, USB Type-A - USB Type-B cable (3 m) x 4, Utility Disk (RadiCS LE,
PDF installation manual), instructions for use

452
476 - 566

Warranty
Dimensions (Unit: mm)Swivel

GX560, GX560-AR: DisplayPort (3 m) x 2

RX560-MD, RX560-AR-MD:
AC power cord (3 m) x 2, USB Type-A - USB Type-B cable (3 m) x 2, Utility Disk
(RadiCS LE, PDF installation manual), instructions for use

452
476 - 566

Video Signals

5MP

5MP5MP

Model Variations

Panel

5MP

6MP

200

308

225

281

200

Please contact the EIZO group company or distributor in your country for the latest information.
Use FDA 510(k) Clearance monitor for diagnosis.
General radiography clearance models do not support display of mammography images for diagnosis.
May vary by country. Please contact EIZO for details.

21

SPECIFICATIONS
Model Variations

2MP

RX370-BK: Anti-Glare coating, withstand, black
RX370-ARBK: Anti-Reflection coating,with stand, black

GX340-CL-BK: Clear Base, with stand,black
GX340-CL-P-BK: Pairing, with stand,black

RX270-BK: Anti-Glare coating, withstand, black
RX270-ARBK: Anti-Reflection coating,with stand, black

GX240-CL-BK: Clear Base, withstand, black
GX240-CL-BK-P: Pairing, with stand,black

MX317W-BK: with stand, black

MX243W-BK: with stand, black

MX217-BK: with stand, black

MX194-BK: with stand, black

MS236WT-ALGY: with Reclining Stand, gray
MS236WT-ALBK: with Reclining Stand, black
MS236WT-AFGY: without stand, gray
MS236WT-AFBK: without stand, black

Type

Color (IPS)

Monochrome (IPS)Color (IPS)

Monochrome (IPS)Color (IPS)Color (IPS)Color (IPS)Color (VA)

Color (IPS)

LED

LED

LED

LED

LED

LED

LED

LED

LED

Size

21.3" (54.1 cm)

21.3" (54 cm)

21.3" (54.0 cm)

21.3" (54 cm)

30.5" (77.5 cm)

24.1" (61 cm)

21.3" (54.0 cm)

19.0" (48.1 cm)

23.0" (58.0 cm)

Native Resolution

1536 x 2048 (3:4 aspect ratio)

1536 x 2048 (3:4 aspect ratio)

1200 x 1600 (3:4 aspect ratio)

1200 x 1600 (3:4 aspect ratio)

4096 x 2160 (17:9 aspect ratio)

1920 x 1200 (16:10 aspect ratio)

1200 x 1600 (3:4 aspect ratio)

1280 x 1024 (5:4 aspect ratio)

1920 x 1080 (16:9 aspect ratio)

Viewable Image Size (H x V)

324.9 x 433.2 mm

324.8 x 433.1 mm

324.0 x 432.0 mm

324.0 x 432.0 mm

685.7 x 361.6 mm

518.4 x 324.0 mm

324.0 x 432.0 mm

376.3 x 301.0 mm

509.2 x 286.4 mm

Pixel Pitch (H x V)

0.2115 x 0.2115 mm

0.2115 x 0.2115 mm

0.270 x 0.270 mm

0.270 x 0.270 mm

0.1674 x 0.1674 mm

0.270 x 0.270 mm

0.270 x 0.270 mm

0.294 x 0.294 mm

0.265 x 0.265 mm

Display Colors / Grayscale Tones

10-bit (DisplayPort): 1.07 billion from apalette of 543 billion (13-bit) colors
8-bit: 16.77 million from a palette of
543 billion (13-bit) colors

10-bit (DisplayPort): 1,024 from apalette of 16,369 (14-bit) tones
8-bit: 256 from a palette of 16,369(14-bit) tones

10-bit (DisplayPort): 1.07 billion from apalette of 543 billion (13-bit) colors
8-bit: 16.77 million from a palette of
543 billion (13-bit) colors

10-bit (DisplayPort): 1,024 from apalette of 16,369 (14-bit) tones
8-bit: 256 from a palette of 16,369(14-bit) tones

10-bit (DisplayPort): 1.07 billion
from a palette of 543 billion (13bit) colors
8-bit: 16.77 million from a palette of
543 billion (13-bit) colors

10-bit (DisplayPort): 1.07 billion from apalette of 543 billion (13-bit) colors
8-bit: 16.77 million from a palette of 543
billion (13-bit) colors

10-bit (DisplayPort): 1.07 billion from apalette of 543 billion (13-bit) colors
8-bit: 16.77 million from a palette of 543
billion (13-bit) colors

10-bit (DisplayPort): 1.07 billion from apalette of 543 billion (13-bit) colors
8-bit: 16.77 million from a palette of 543
billion (13-bit) colors

8-bit: 16.77 million from a palette of 1.06
billion (10-bit) colors

Viewing Angles (H / V, typical)

178° / 178°

176° / 176°

178° / 178°

176° / 176°

178° / 178°

178° / 178°

178° / 178°

178° / 178°

178° / 178°

Brightness (typical)

1100 cd/m²

1200 cd/m²

1000 cd/m²

1200 cd/m²

550 cd/m²

410 cd/m²

500 cd/m²

350 cd/m²

260 cd/m²

Recommended Brightness for
Calibration

500 cd/m²

500 cd/m²

500 cd/m²

500 cd/m²

270 cd/m²

220 cd/m²―――

Contrast Ratio (typical)

1800:1

1400:1

1800:1

1400:1

1800:11350:1

1800:1

2000:1

1000:1

Response Time (typical)

25 ms (black-white-black)40 ms (black-white-black)

20 ms (black-white-black)40 ms (black-white-black)25 ms (black-white-black)22 ms (black-white-black)

20 ms (black-white-black)

20 ms (black-white-black)11 ms (gray-to-gray)

Type

―

―

―

―

―

―

―

―

Projected Capacitive

Touch Points

―

―

―

―

―

―

―

―

10

Surface Treatment

―

―

―

―

―

―

―

―

Anti-Glare coating

Communication Protocol

―

―

―

―

―

―

―

―

USB

Surface Hardness

―

―

―

―

―

―

―

―

5H

Compatible OS

―

―

―

―

―

―

―

―

Windows 11, Windows 10 (64-bit, 32-bit)

Input Terminals

DisplayPort x 2, DVI-D (dual link)

DisplayPort, DVI-D (dual link)

DisplayPort x 2, DVI-D

DisplayPort, DVI-D

USB Type-C (DisplayPort Alt Mode),
DisplayPort x 2, HDMI

DisplayPort, DVI-D

DisplayPort, DVI-D

DisplayPort, DVI-D, D-Sub mini 15 pin

DisplayPort (HDCP1.3), HDMI (HDCP1.4),
D-Sub mini 15 pin

Output Terminals

DisplayPort (daisy chain)―

DisplayPort (daisy chain)―

USB Type-C (daisy chain)

DisplayPort (daisy chain)

DisplayPort (daisy chain)――

Digital Scanning Frequency (H / V)

31 - 127 kHz / 29 - 61.5 Hz

31 - 127 kHz / 29 - 61.5 Hz

31 - 100 kHz / 59 - 61 Hz

31 - 100 kHz / 59 - 61 Hz

USB Type-C: 31 - 134 kHz / 59 - 61 Hz
DisplayPort: 31 - 134 kHz / 59 - 61 Hz
HDMI: 31 - 136 kHz / 59 - 61 Hz

31 - 76 kHz / 59 - 61 Hz

31 - 100 kHz / 59 - 61 Hz

31 - 64 kHz / 59 - 61 Hz

31 - 68 kHz / 59 - 61 Hz

Analog Scanning Frequency (H / V)

―

―

―

―

―

―

―

24.8 - 80 kHz / 50 - 75 Hz

31 - 81 kHz / 55 - 76 Hz

Sync Formats

―

―

―

―

―

―

―

Separate

Separate

UpstreamUSB 2.0: Type-B x 2

USB 2.0: Type-B

USB 2.0: Type-B x 2

USB 2.0: Type-B

USB 2.0: Type-B x 2,
USB 2.0: Type-C (DisplayPort Alt Mode,
Power Delivery Source 94 W max.)

USB 2.0: Type-B

USB 2.0: Type-B

USB 2.0: Type-B

USB 5Gbps: Type-B

Downstream

USB 2.0: Type-A x 2

USB 2.0: Type-A x 2

USB 2.0: Type-A x 2

USB 2.0: Type-A x 2

USB 2.0: Type-A x 3
USB 2.0: Type-C (DisplayPort Alt Mode,
Power Delivery Source 15 W max.)

USB 2.0: Type-A x 2

USB 2.0: Type-A x 2―

USB 5Gbps: Type-A x 2―

USB LAN Adapter

―

―

―

―

RJ-45 (1000BASE-T)

―

―

―

Dedicated Charging PortUSB Type-C (Power Supply 15 W max.)―

USB Type-C (Power Supply 15 W max.)

―

―

―

―

―

―

Power Input

AC 100 - 240 V: 50 / 60 Hz

AC 100 - 120 V, 200 - 240 V: 50 / 60 Hz

AC 100 - 240 V: 50 / 60 Hz

AC 100 - 120 V, 200 - 240 V: 50 / 60 Hz

AC 100 - 240 V: 50 / 60 Hz

AC 100 - 240 V: 50 / 60 Hz

AC 100 - 240 V: 50 / 60 Hz

AC 100 - 240 V: 50 / 60 Hz

AC 100 - 240 V: 50 / 60 Hz

Typical Power Consumption

36 W

36 W

33 W

29 W

62 W

26 W

23 W

15 W

15 W

Maximum Power Consumption

105 W

90 W

98 W

76 W

260 W

56 W

54 W

28 W

47 W

Power Save Mode1 W or less

1.6 W or less

1 W or less

1.6 W or less

0.5 W or less

0.6 W or less

0.6 W or less

0.6 W or less

0.5 W or less

Backlight Sensor, Integrated Front
Sensor, Ambient Light Sensor

Backlight Sensor, Integrated Front
Sensor, Presence Sensor, Ambient
Light Sensor

Backlight Sensor, Integrated Front
Sensor, Ambient Light Sensor

Backlight Sensor, Integrated Front
Sensor, Presence Sensor, Ambient
Light Sensor

Backlight Sensor, Integrated Front
Sensor, Ambient Light Sensor

Backlight Sensor

Backlight Sensor, Integrated Front Sensor,
Ambient Light Sensor

Backlight Sensor―

Brightness Stabilization

Yes

Yes

Yes

Yes

Yes

Yes

Yes

Yes

―

Digital Uniformity Equalizer

Yes

Yes

Yes

Yes

Yes

Yes

Yes

Yes

―

Hybrid Gamma PXL

Yes

―

Yes

―

Yes

Yes

Yes

―

―

Work-and-Flow

Hide-and-Seek, Switch-and-Go, Pointand-Focus, Instant Backlight Booster―

Switch-and-Go, Point-and-Focus,
Instant Backlight Booster―

Hide-and-Seek, Switch-and-Go,
Point-and-Focus

Point-and-Focus

Point-and-Focus――

Preset Modes

CAL Switch (DICOM, CAL1, CAL2,
Custom, sRGB, Text)

CAL Switch (DICOM, CAL1, CAL2,
Hybrid-γ)

CAL Switch (DICOM, CAL1, CAL2,
Custom, sRGB, Text)

CAL Switch (DICOM, CAL1, CAL2,
Hybrid-γ)

CAL Switch (DICOM, CAL, Patho,
Custom, sRGB, Text)

CAL Switch (DICOM, CAL1, CAL2, Custom,sRGB, Text)

CAL Switch (DICOM, CAL1, CAL2, Custom,sRGB, Text)

CAL Switch (DICOM, CAL1, CAL2, Custom,sRGB, Text)

User1, User2, sRGB, DICOM

OSD Languages

English, German, French, Italian,Japanese, Simplified Chinese, Spanish,
Swedish, Traditional ChineseEnglish, German, French, Italian,Japanese, Simplified Chinese, Spanish,
Swedish, Traditional ChineseEnglish, German, French, Italian,Japanese, Simplified Chinese, Spanish,
Swedish, Traditional ChineseEnglish, German, French, Italian,Japanese, Simplified Chinese, Spanish,
Swedish, Traditional ChineseEnglish, German, French, Italian,Japanese, Simplified Chinese,
Spanish, Swedish, Traditional ChineseEnglish, German, French, Italian, Japanese,
Simplified Chinese, Spanish, Swedish,Traditional ChineseEnglish, German, French, Italian, Japanese,
Simplified Chinese, Spanish, Swedish,Traditional ChineseEnglish, German, French, Italian, Japanese,
Simplified Chinese, Spanish, Swedish,Traditional ChineseEnglish, German, French, Italian, Japanese,
Simplified Chinese, Spanish, Swedish,Traditional Chinese

Net Weight

8 kg

10.2 kg

7.7 kg10.2 kg

12.4 kg

7.8 kg

7.2 kg6 kg

6.6 kgNet Weight (Without Stand)5.2 kg

7.5 kg

4.9 kg

7.5 kg

8.2 kg

4.9 kg

4.3 kg

4.2 kg

6 kg

Hole Spacing (VESA Standard)

100 x 100 mm

100 x 100 mm

100 x 100 mm

100 x 100 mm

100 x 100 mm

100 x 100 mm

100 x 100 mm

100 x 100 mm

100 x 100 mm

CE / UKCA (Medical Device), ANSI/AAMI
ES60601-1, CAN/CSA-C22.2 No. 606011, IEC/EN60601-1, VCCI-B, FCC-B, CAN
ICES-3 (B), RCM, RoHS, China RoHS,
WEEE, CCC, EAC

CE (Medical Device), EN60601-1, ANSI/AAMI ES60601-1, CAN/CSA-C22.2 No.
60601-1, IEC60601-1, VCCI-B, FCC-B,
CAN ICES-3 (B), RCM, RoHS, China
RoHS, WEEE, CCC, EAC

CB, CE / UKCA (Medical Device), ANSI/AAMI ES60601-1, CAN/CSA-C22.2
No. 60601-1, IEC/EN60601-1, VCCI-B,
FCC-B, CAN ICES-3 (B), RCM, RoHS,
China RoHS, WEEE, CCC

CE (Medical Device), EN606011, ANSI/AAMI ES60601-1, CAN/CSA-C22.2 No. 60601-1, IEC60601-1,
VCCI-B, FCC-B, CAN ICES-3 (B), RCM,
RoHS, China RoHS, WEEE, CCC, EAC

CB, CE / UKCA (Medical Device), ANSI/AAMI ES60601-1, CAN/CSA-C22.2
No. 60601-1, IEC/EN60601-1, VCCI-B,
FCC-B, CAN ICES-3 (B), RCM, RoHS,
China RoHS, WEEE, CCC

CE / UKCA (Medical Device), ANSI/AAMI
ES60601-1, CAN/CSA-C22.2 No. 60601-1, IEC/
EN60601-1, VCCI-B, FCC-B, CAN ICES-3 (B),RCM, RoHS, China RoHS, WEEE, CCC, EAC

CB, CE / UKCA (Medical Device), ANSI/AAMI
ES60601-1, CAN/CSA-C22.2 No. 60601-1, IEC/
EN60601-1, VCCI-B, FCC-B, CAN ICES-3 (B),RCM, RoHS, China RoHS, WEEE, CCC

CE / UKCA (Medical Device), ANSI/AAMI
ES60601-1, CAN/CSA-C22.2 No. 60601-1, IEC/
EN60601-1, VCCI-B, FCC-B, CAN ICES-3 (B),RCM, RoHS, China RoHS, WEEE, CCC, EAC

CE / UKCA (Medical Device), ANSI/AAMI
ES60601-1, CAN/CSA-C22.2 No. 60601-1, IEC/
EN60601-1, PSE, VCCI-B, FCC-B, CAN ICES-3 (B),RCM, RoHS, China RoHS, WEEE, CCC

Certifications & Standards 1

FDA 1, 2, 3
Monitor Quality Control Software RadiCS

510(k) Clearance for General
Radiography

510(k) Clearance for General
Radiography

510(k) Clearance for General
Radiography

510(k) Clearance for General
Radiography

Pending: 510(k) Clearance forGeneral Radiography

510(k) Clearance for General Radiography

510(k) Clearance for General Radiography

510(k) Clearance for General Radiography

Class I

Supported

Supported

Supported

Supported

Supported

Supported

Supported

Supported―

Signal Cables

DisplayPort (3 m) x 2

Dual Link DVI-D (3 m), DisplayPort(3 m)

DisplayPort (3 m) x 2

DVI-D (3 m), DisplayPort (3 m)

USB Type-C (1.5 m), DisplayPort (3m) x 2, HDMI (3 m)

DisplayPort (3m)

DisplayPort (3 m)

DisplayPort (3 m)

DisplayPort (3 m), HDMI (3 m)OthersAC power cord (3 m), USB Type-A USB Type-B cable (3 m) x 2, Utility Disk
(RadiCS LE, PDF instructions for use,
PDF installation manual), instructions
for useAC power cord (3 m), USB Type-A USB Type-B cable (3 m), Utility Disk
(RadiCS LE, user's manual)

AC power cord (3 m), USB Type-A USB Type-B cable (3 m) x 2, Utility Disk
(RadiCS LE, PDF instructions for use,
PDF installation manual), instructions
for useAC power cord (3 m), USB Type-A USB Type-B cable (3 m), Utility Disk
(RadiCS LE, user's manual)

AC power cord (3 m), USB Type-A USB Type-B cable (3 m) x 2, Utility
Disk (RadiCS LE, PDF installation
manual), instructions for use

AC power cord (3 m), USB Type-A - USB
Type-B cable (3m), Utility Disk (RadiCS LE,
PDF instructions for use, PDF installation
manual), instructions for use

AC power cord (3 m), USB Type-A - USB
Type-B cable (3 m), Utility Disk (RadiCS LE,
PDF instructions for use, PDF installation
manual), instructions for use

AC power cord (3 m), USB Type-A - USB
Type-B cable (3 m), Utility Disk (RadiCSLE, PDF installation manual), instructions
for useAC power cord (3 m), USB Type-A - USB
Type-B cable (3 m), touch pen, holder for
touch pen, Utility Disk (user's manual, touch
panel driver, TPOffset), cleaning cloth, mask
sheet x 2 (MS236WT-AL), Screw for VESA
mount x 4 (MS236WT-AF), Cable clamper

5 Years

5 Years

5 Years

5 Years

5 Years

5 Years

5 Years

5 Years

3 Years

RX370, RX270, MX317W,
MX243W, MX217

GX340, GX240

MX194

Swivel not supported with MS236WT-A.

22

281

200

211

245.5

Please contact the EIZO group company or distributor in your country for the latest information.
Use FDA 510(k) Clearance monitor for diagnosis.
General radiography clearance models do not support display of mammography images for diagnosis.
May vary by country. Please contact EIZO for details.

281

90°

200

211

5° 30°
73

721

245.5

308.1

225.1

5° 30°
64.5

552

281

200

5°

356.5

281

30°
70.5

90°

200

320

70

°

205

216.3
54

556.7

30°
61.5

405

15°

353.9

376

339.2
360.7

90°

78

401.3
143.9

70°

356.5

334
406.5 - 506.5

344°

376

30°
98

464.5
482.3 - 572.3

70°

463
481.5 - 571.5

90°

1
2
3
4

5° 30°
78341.3

5° 30°°

Swivel30°
9890

Dimensions (Unit: mm)

361.6
430.8 - 520.8

Warranty

90°

Dedicated
SoftwareSupplied
Accessories 4

1MP

401
469.5 - 569.5

PhysicalSpecifications

2MP

Backlight

SensorFeatures &Functions

2.3MP

505
520 - 599

Power

8MP

464.5
482.3 - 572.3

USB

2MP

°

Video Signals

2MP

90

Touch Panel

3MP

505
520 - 599

Panel

3MP

23

GRAPHICS BOARDSMONITOR QUALITY CONTROL SOLUTIONS

To get the most out of the extraordinary capabilities of our

Monitor Quality Control
Software & Calibration Sensor

high-definition RadiForce monitors, we recommend thatyou use them with one of EIZO's dedicated graphics boards.
Compatible Operating Systems

monitor solutions and achieves the native resolution and highWindows 11
Windows 10
macOS Ventura (13)macOS Monterey (12)Display Functions

performance required for making precise diagnoses.

DICOM Part 14 GSDF, CIE, Exponential(gamma value), Log Linear, Linear, User
definition

InterfaceUSB, RS232C (Windows only)

Languages

English, German, Japanese, Chinese,
French, Spanish

Package Contents

RadiCS DVD-ROM (RadiCS, User's Manual),
UX2 Sensor, Adsorptive sheet for the
replacement, cleaning cloth, UX2 Sensor
Instructions for Use

Each board is used to specifically support RadiForce medical

Software for major version up of RadiCS.
Bus Interface

PCI-Express x16

PCI-Express x16

PCI-Express x16

PCI-Express x16

Compatible OS

Windows 10

Windows 10

Windows 10 / 8.1 / 7

Windows 10 / 8.1 / 7

Memory

8 GB

5 GB

4 GB

2 GB

Display Colors / Grayscale Tones

10-bit (DisplayPort, USB Type-C),
8-bit

10-bit (DisplayPort), 8-bit

10-bit (DisplayPort), 8-bit

10-bit (DisplayPort), 8-bit

Output Terminals

DisplayPort x 3(Daisy chain supported), USB Type-C x 1

DisplayPort x 4(Daisy chain supported)

Mini DisplayPort x 4(Daisy chain supported)

Mini DisplayPort x 3(Daisy chain supported)

Supplied Conversion Cables

DisplayPort - DVI-D

DisplayPort - DVI-D

Mini DisplayPort - DisplayPort x 2,
Mini DisplayPort - DVI-D

Mini DisplayPort - DisplayPort x 2,
Mini DisplayPort - DVI-D

Maximum Connected Monitors

Four Monitors

Four Monitors

Four Monitors

Four Monitors

75 W

47 W

30 W

Maximum Power Consumption

125 W (not using USB Type-C power
delivery)
160 W (using USB Type-C power
delivery)Slot (s)

1

1

1

1

ChassisStandard

Standard

Standard & Low-Profile

Standard & Low-Profile

Dimensions (W x H)

241.3 x 104.9 mm

200.1 x 111.1 mm

150.0 x 68.9 mm

149.9 x 68.9 mm

RX1270Recommended

Yes

Yes

Yes

RX660

Recommended

Yes

Yes

Yes

RX560

Recommended

Yes

Yes

Yes

GX560

Recommended

Yes

Yes

Yes

RX370

Yes

Recommended

Yes

GX340

Yes

Recommended

RX270

Yes

GX240

Yes

Network QC
Management Software

Manageable Number of PCs /Monitors

1000 PCs / 8000 Monitors Maximum

Administrator PC Browser

Microsoft Edge 79 or later
Google Chrome™ browser

Administrator PC Resolution

1024 x 768 MinimumServer PC Operating Systems

Windows Server 2022 Standard
Windows Server 2019 Standard
Windows Server 2016 Standard
Windows Server 2012 R2 Standard
Windows 11 Pro / Enterprise
Windows 10 Pro / Enterprise (64-bit)

Server PC Database

SQL Server 2022 Standard / Express
EditionSQL Server 2019 Standard/Express
EditionSQL Server 2016 Standard / Express
Edition SP2Server PC Hard Disk Drive

150 GB MinimumServer PC Memory

8 GB Minimum

Languages

English, German, Japanese, Chinese,
French, Spanish

Monitor Access License must be purchased
for every 5 additional monitors when using
RadiNET Pro Version 5.

A license to use RadiCS with othercommercially available monitors.

ACCESSORY
Care for the Radiologist's Eyes

Comfort Light for Reading Rooms

Relief with Gentle Light
Cabinet Color

Black

Power Requirements

USB power

Weight370 g

Yes

Dimensions (W x H x D)

184 x 185.5 x 15.7 mm

Yes

Yes

Certifications & Standards

CE, UKCA, IEC60950-1, CSA C22.2 No.
60950-1, VCCI-B, FCC-B, CAN ICES-3 (B),RCM, RoHS, China RoHS, WEEE

Yes

Yes

Recommended

Supplied Accessories

Yes

Recommended

Yes

dedicated cable, user's manual,
mounting bracket, spacers, screws

Warranty

3 Years

MX317W

Yes

Yes

Recommended

Yes

MX243W

Yes

Yes

Yes

Yes

MX217

Yes

Yes

Yes

Recommended

MX194

Yes

Yes

Yes

Recommended

MS236WT-A

Yes

Yes

Yes

Recommended

RadiLight attaches to the back of RadiForce monitorsand shines a light on the wall behind it. This eases
the amount of concentrated light traveling to theradiologist's eyes for reducing eyestrain while notimpacting the reading room's overall ambient lighting
or visibility of the images on the screen.

Flicker-Free
RadiLight is a flicker-free lighting
solution that reduces eyestrain.

Spotlight
RadiLight Focus allows you
to check or read printed
documents or see yourkeyboard and other tools.

Graphics board compatibility is subject to change without notice. Please check EIZO website for updates.

Easily Attachable
RadiLight easily attaches to the backof the monitor stand so it does nottake up desk space.

The brightness can be adjusted to 10 different levels.

24

25

E

xt

Built-In Color Calibration Sensors

ova
Inn

5 0 + Ye a

rs o

fE

a ch

IP Decoding Solutions

VisualTechnology
Company

Re

t iv

et

eS

ark

olu

eM

tio

s iv

ns

en

Automatically calibrates while you work

Business Enterprise

Creative Work

x
pe

r ti

LAN CablesePC-less connection

SwitchVisibility Enhancement Technology／

Healthcare
Ultra-High Sensitivity
HD Cameras

Global Reach

Research andDevelopment

Image Optimization System

Market-Focused Cloud Solutions
Air Traffic Control

Creative Work

Manufacturing

Healthcare

Security & Surveillance /Maritime

Customization

Quality Control

Software for Improved Workflow／／

Air Traffic Control
26

Use a single mouse across two PCs

Synchronized adjustment of multiple monitors

Simplified CMS with automatic software andprinter settings adjustment27

153 Shimokashiwano, Hakusan, Ishikawa 924-8566 Japan
Phone: +81-76-277-6794

https://www.eizoglobal.com
EIZO, the EIZO Logo, ColorEdge, CuratOR, DuraVision, FlexScan, RadiCS, RadiForce, RadiNET, and Raptor are registered trademarks of EIZO Corporation in
Japan and other countries. RadiLight, Re/Vue, SafeGuard, and ScreenCleaner are trademarks of EIZO Corporation. The terms HDMI, HDMI High-Definition
Multimedia Interface, HDMI trade dress and the HDMI Logos are trademarks or registered trademarks of HDMI Licensing Administrator, Inc. DisplayPort, theDisplayPort Compliance Logo and VESA are trademarks of the Video Electronics Standards Association in the United States and other countries. USB Type-C is a
registered trademark of USB Implementers Forum, Inc. DICOM is the registered trademark of the National Electrical Manufacturers Association for its standards
publications relating to digital communications of medical information. Microsoft, Microsoft Edge, SQL Server, Windows, and Windows Server are registeredtrademarks of Microsoft Corporation in the United States and other countries. macOS is registered trademarks of Apple Inc. All other company and productnames and logos are trademarks or registered trademarks of their respective owners. Specifications are subject to change without notice.

Copyright © 2023 EIZO Corporation. All rights reserved. (231003)