LifeStyle™ Portable
Oxygen Concentrator
Service Manual

AirSep Corporation • 401 Creekside Drive • Buffalo, NY 14228-2085 USA
Telephone: 716. 691.0202 • 24-Hour Fax: 716.691.4141 • www.airsep.com

Expedited Product Warranty Check service is always at your fingertips with AirSep:
Type www.airsep.info/warranty_check/main.asp
OR
In the US or Canada,dial 866-873-9277

AirSep® is a registered trademark of AirSep Corporation.


LifeStyle™ is a trademark of AirSep Corporation.

MN110-1

Rev A

07/07

Table of Contents

Section 1.0
Introduction
1.1

Equipment Provider Responsibility

1-1

1.2

Important Notice and Symbol Explanations

1-2

1.3

Definitions of Symbols

1-3

1.4

LifeStyle Product Limited Warranty

1-4

1.5

LifeStyle Specifications

1-5

1.6

LifeStyle Classifications

1-6

Section 2.0
Operation Check andOxygen Concentration Test
2.1

Description of Operation

2-1

2.2

Operation Check

2-1

2.3

Alarm System

2-2

2.3.1

Start-up

2-2

2.3.2

Low Battery

2-2

2.3.3

Discharged Battery

2-2

2.3.4

Battery Charging

2-2

2.3.5

Battery Fully Charged

2-3

2.3.6

Cannula Disconnect

2-3

2.3.7

Performance Warning

2-3

2.3.8

LifeStyle Capacity Exceeded

2-3

2.3.9

General Malfunction

2-4

2.4

Oxygen Concentration Test and Specification

2-4

3.1

General Instructions

3-1

3.2

Routine Maintenance by the Patient

3-1

Section 3.0
Patient Instructions

3.2.1
3.2.2
3.3
AirSep Corporation

Cleaning the Air Intake
Gross Particle Filter

3-1

Checking the Hour Meter

3-2

Usage at Night or with Extended Tubing

3-2
i – table of contents

Section 4.0
Equipment Provider
Maintenance
4.1

Routine Maintenance

4-1

4.1.1

Air Intake / Gross Particle Filter

4-1

4.1.2

Product Filter Replacement

4-1

4.1.3

Compressor Replacement

4-2

4.1.4

Recording Maintenance

4-2

4.2

Infection Control

4-2

4.3

Cleaning and Preparing for New Patient Use

4-2

5.1

Components

5-1

5.2

Housing Removal

5-1

5.2.1

Removing Bottom Housing

5-1

5.2.2

Removing Top Housing

5-2

Section 5.0
Service

5.3

Compressor Replacement

5-2

5.3.1

Compressor Assembly Removal

5-2

5.3.2

Compressor Assembly Installation

5-3

Section 6.0
Troubleshooting
6.1

General Troubleshooting

6-1

6.2

Troubleshooting Chart

6-2

Figure 1: LifeStyle — Control Panel

A-1

Figure 2: LifeStyle — Internal, Top View

A-2

Figure 3: LifeStyle — Internal, Bottom View

A-3

Appendix
Exploded Drawings

ii – table of contents

AirSep Corporation

1.0

Introduction
1.1

Equipment Provider Responsibility
All Equipment Providers of the LifeStyle™ Portable Oxygen Concentrator must assume
responsibilities for handling, operational check-out, patient instruction, and maintenance.
These responsibilities are outlined below and throughout this manual.
WARNING
This unit is not a life support device. Geriatric, pediatric,
and other patients unable to communicate discomfort while
using this oxygen concentrator may require additional
monitoring. Patients with hearing and/or sight impairment(s)may need assistance with monitoring the alarms.Instructpatients to consult their physician immediately if theyexperience any signs of discomfort.

As an Equipment Provider, you must do all of the following:Inspect the condition of each LifeStyle unit immediately upon delivery to yourbusiness location. Note any sign of damage, external or internal, on thedelivery receipt, and report it directly to both the freight company and AirSep
Corporation immediately.



Check the operation of each LifeStyle unit before delivery to a patient.
Always operate the unit for a reasonable length of time, and check that theoxygen concentration level is within specifications, as described in Section 2.4of this manual.

Charge the battery initially for 12 hours. See “Battery Charging” in Section
2.3.4 of this manual.

Deliver LifeStyle units only to patients authorized by a physician’s prescription.



Instruct patients how to use LifeStyle in conjunction with the Patient Manual.

1-1

1.2



Instruct patients to notify their physicians and/or Equipment Providers if theyexperience any signs of discomfort.



Instruct each patient how to perform routine maintenance of the air intake
gross particle filter and how to monitor the hour meter for the compressor.
(Refer to Section 3.2.)



Be available to service each patient at any time.



Maintain LifeStyle in accordance with Section 4.0.

Repair components and replace parts only as outlined in this manual. Use
only AirSep parts for replacement in LifeStyle Oxygen Concentrators.



Refer to the “LifeStyle Product Limited Warranty,” Section 1.4 of this manual,
if replacement of parts is required within the warranty period.

Important Notice and Symbol Explanations
As you review both the LifeStyle Patient Manual and this LifeStyle Service Manual,
pay special attention to the WARNING, CAUTION, and NOTE messages. They
identify safety guidelines or other important information as follows:

WARNING:

Describes a hazard or unsafe practice that can result in severe
bodily injury or death and/or property damage.

CAUTION:

Describes a hazard or unsafe practice that can result in minor
bodily injury or property damage.

NOTE:

Provides information important enough to emphasize or repeat.

1-2

1.3

Definitions of Symbols

The definitions of symbols used with the LifeStyle Portable Oxygen Concentrator follow.

SymbolDescription
Refer to Patient Manual instructions

No oilNo smoking

Type BF equipment

Class IICE Mark
Safety agency for CAN/CSA C22.2
No. 601.1.M90 for medical electricalequipment.

FAA SFAR106

Proper disposal of waste of electricaland electronic equipment required.

1-3

1.4

LifeStyle Product Limited Warranty
AirSep Corporation warrants the LifeStyle Oxygen Concentrator to be free fromdefect in parts for two years (or as specified on the original invoice provided) from thedate of delivery to the original purchaser, under normal use and operation. The battery is
warrantied for 90 days. AirSep Corporation’s obligations under this warranty are limited
to the repair or replacement of any such item of equipment (or part thereof) shown to
be defective or, at AirSep Corporation’s option, to refund the purchase price of any
such defective item of equipment.
Each item of equipment for which a warranty claim is asserted shall, at the requestof AirSep Corporation, be returned on a prepaid basis with proof of purchase date to
the AirSep factory at the expense of the purchaser. The purchaser will be responsible
for return freight charges. Replacement parts shall be warranted as stated above for theunexpired portion of the original two-year parts warranty (as specified on the original
invoice provided). This warranty does not extend to any item or part subjected tomisuse, accident, improper maintenance, or application, or which has been repaired oraltered outside of the AirSep Corporation factory without the express prior written
authorization of AirSep Corporation.
THE FOREGOING WARRANTY IS IN LIEU OF ANY OTHER WARRANTY, EXPRESSED OR IMPLIED, IN FACT OR IN LAW, INCLUDINGWITHOUT LIMITATION THE WARRANTY OF MERCHANTABILITY OR THE
WARRANTY OF FITNESS FOR PARTICULAR PURPOSE. IT IS EXPRESSLYUNDERSTOOD THAT PURCHASER’S SOLE AND EXCLUSIVE REMEDY FOR
DEFECT IN PARTS IS LIMITED TO ENFORCEMENT OF AIRSEPCORPORATION’S OBLIGATION AS SET FORTH ABOVE, AND AIRSEPCORPORATION SHALL NOT BE LIABLE TO PURCHASER OR OTHERS FOR
LOSS OF USE OF THE EQUIPMENT OR FOR OTHER SPECIAL, INDIRECT,INCIDENTAL, OR CONSEQUENTIAL DAMAGES.

1-4

1.5

LifeStyle Specifications
Oxygen Concentration:*Set at 1-5 LPM: equivalent to a continuous flow of 90%
oxygen ± 3%

Dimensions:

5.5 in. high x 7.25 in. wide x 16.31 in. long
(13.9 cm high x 18.4 cm wide x 41.4 cm long)

Weight:9.75 lb (4.4 kg)

Power:LifeStyle unit - 13 VDC, 42 watts
Power supply - 100 – 250 VAC, 50/60 Hz, 2.0 amps max
Battery pack - 12 VDC, nominal

Battery duration:

50 minutes nominal for fully-charged battery

Battery charge time:2 — 2½ hours

Warm-up time:

3 minutes

Audible alarms andvisual indicators:

Start-up – audible and visual (GREEN indicator light)
Pulse flow – visual (GREEN pulse light)
Battery condition – audible and visual (YELLOW indicator,
warning, and alarm lights)
Cannula disconnect - audible and visual (RED alarm light)
System overdraw – audible and visual (RED warning andalarm lights)
High and low pressure – audible and visual (RED alarm light)

Sound:

Less than 55 decibels

Temperature range
intended for use:

Operational temperature: 41oF to 104oF (5oC to 40oC)
Storage temperature:-4oF to 140oF (-20oC to 60oC)* Based on an atmospheric pressure of 14.7 psia (101 kPa) at 70oF (21oC)

1-5

1.6

LifeStyle Classifications
Type of protection against electric
shock:Class II

Protection from electric shock is achieved bydouble insulation.

Degree of protection againstelectric shock:Type BF

Equipment providing a particular degree ofprotection against electric shock regarding1) allowable leakage current2) reliability of protective earth connection(if present).
Not intended for direct cardiac application.

Independent testing for
Medical Electrical Equipment
Standard

Tested by UltraTech Group of Labs to be incompliance with IEC 60601-1:1988 +A1:1991, A2:1995

Protection against potential
electromagnetic or otherinterference between theequipment and other devices:

Tested by UltraTech Group of Labs to be incompliance with IEC 601-1-2 (EMC)

Method of cleaning andinfection control allowed:

Please refer to “Provider Maintenance,”
Section 4.0 of this LifeStyle Service Manual.

Degree of safety of
application in the presenceof flammable anesthetic
gases:

Equipment not suitable for such application.

Mode of operation:

Continuous duty

FAA SFAR 106:

RTCA DO160D, Section 21,
Category M-Compliant

1-6

2.0

Operation Check and Oxygen Concentration Test
2.1 Description of Operation
Air enters the LifeStyle Oxygen Concentrator through an external air intake / grossparticle filter. This filtered air enters the compressor via a suction resonator that quiets
the compressor’s suction sound. Pressurized air then exits the compressor and passes
through a heat exchanger. The heat exchanger reduces the temperature of the compressed air. Next, a two-way solenoid feed valve directs the air into one of two sieve
beds that contain molecular sieve. The unique property of molecular sieve enables it tophysically attract (adsorb) nitrogen when air passes through this material, thereby
separating the nitrogen from the oxygen in ambient air.
There are two sieve beds: While one produces high-concentration oxygen, the other ispurged of the nitrogen it adsorbed (collected) while it made oxygen. Each adsorber(bed) produces oxygen for approximately eight seconds and delivers it through a
product tank to the regulator. From the regulator, oxygen passes through the productfilter and then to the outlet fitting.
In addition to the molecular sieve beds that allow oxygen to be separated from air,
LifeStyle contains an Oxygen Conserving Device (OCD) valve that controls delivery ofhigh-concentration oxygen to the patient at the point in the breathing cycle when thepatient starts to inhale.
LifeStyle delivers a gas consisting of 90% (±3%) oxygen at pulse flow rates from 1–5 lpm.

2.2 Operation Check
AirSep tests every LifeStyle Oxygen Concentrator thoroughly after manufacture. You
must perform the following test to ensure that no damage occurred in shipping or
handling.
1.

Open and inspect all cartons upon delivery. Unpack the unit and remove itfrom the carton. Inspect the unit itself for damage. If the exterior of a unit’s
carton is damaged, or if the unit itself is damaged, note this on the freight billsigned by the driver.

2.

Connect the power cord of the unit, and set the ON/OFF switch to the ON
position. Check to see that the following sequence occurs:

3.

a.

A brief, continuous alarm sounds to indicate that LifeStyle has
been turned on.

b.

The compressor runs.

c.

Exhaust air flows from the unit.

Perform an oxygen concentration test, as described in Section 2.4.
2-1

2.3

Alarm System
An audible alarm sounds if LifeStyle has a low battery, senses no inspiration, or if
performance outside of specifications is detected. If an alarm sounds, open the cover,
and observe which lights are on. The visual and audible alarm conditions are explained
in detail below.

2.3.1

Start-Up

A five-second, continuous, audible alarm sounds to indicate start-up. LifeStyle
begins to operate when the alarm stops.

2.3.2

Low Battery

As the battery power approaches a low level, a one-second beep is emitted every
five seconds, and the yellow BATTERY light flashes for one second every fiveseconds. When this occurs, the patient is instructed to replace the battery, connect
LifeStyle to a DC auto adapter power outlet or an AC power outlet, or change toanother source of oxygen within five minutes. When LifeStyle is connected to a DC
auto adapter outlet or to an AC outlet, power from the automobile battery or fromthe AC outlet operates LifeStyle and recharges LifeStyle’s battery simultaneously.

2.3.3

Discharged Battery

LifeStyle does not produce oxygen when the battery power falls below a certain
level. When this occurs, a half-second beep sounds every half second, and the
YELLOW battery light flashes. In this case, the patient is instructed to replace thebattery, connect LifeStyle to a DC auto adapter outlet or an AC outlet, or change toanother source of oxygen immediately. When LifeStyle is connected to a DC auto
adapter or an AC outlet, power from the automobile battery or from the AC outlet
operates LifeStyle and recharges LifeStyle’s battery simultaneously.

2.3.4

Battery Charging

Charge the battery for the first time for a minimum of 12 hours. The BATTERY lightmay illuminate a steady yellow significantly before 12 hours; however, the battery
should be left in LifeStyle with LifeStyle connected to the DC auto adapter or AC
adapter for the full 12 hours. When LifeStyle is plugged into an automobile’s DC
outlet or an AC outlet and is turned OFF, the yellow BATTERY light flashes toindicate that the battery is being charged. When LifeStyle is turned ON and plugged
into the DC auto adapter or AC adapter, the battery still charges; however, theyellow BATTERY light is not on.

2-2

2.3.5

Battery Fully Charged

When LifeStyle is connected to an AC or DC outlet and is in the OFF setting, theyellow BATTERY light illuminates and displays continuously to indicate that thebattery is fully charged. However, when charging the battery for the first time, thebattery should be left in the charger for 12 hours even if the yellow BATTERY lightis on continuously.

2.3.6

Cannula Disconnect

When LifeStyle is connected to a power source and does not sense breathing withina predetermined time period, a constant alarm sounds, and the red ALARM lightilluminates. If this occurs, the patient is instructed to check the connection from thecannula to the LifeStyle unit, make sure that the nasal cannula is positioned properlyon the face, and ensure that breathing is through the nose. If the alarm continues tosound, the patient should change to an alternate source of oxygen and contact theEquipment Provider.

2.3.7

Performance Warning

The performance of LifeStyle can be affected by two conditions. The first is if theuser’s breathing rate causes the LifeStyle unit’s capacity to be exceeded. The
LifeStyle unit supplies a fixed amount of high-concentration oxygen. The amount ofoxygen that LifeStyle supplies is dependent on the flow control switch setting andthe breathing rate. If the breathing rate is too fast for the flow selector setting, thecapacity of LifeStyle can be exceeded.
The second condition that can result in decreased performance is a generalmalfunction. LifeStyle monitors its performance and alerts the patient if this occurs.
If either of these events occurs, a half-second alarm sounds every five seconds. If
breathing is too fast for the flow selector setting and LifeStyle capacity is exceeded,
the red ALARM light flashes. If a general malfunction occurs, a steady red
ALARM light appears. During this time, LifeStyle is still producing high-purity
oxygen and may continue to be used. The patient should reduce activity or consider
locating another source of oxygen.

2.3.8

LifeStyle Capacity Exceeded

When the breathing rate has caused the capacity of LifeStyle to be exceeded, ahalf-second alarm sounds every half second, and the red alarm light flashes. When
this occurs, oxygen concentration is outside specifications. The patient shouldchange to another source of oxygen and contact the Equipment Provider.

2-3

2.3.9General Malfunction

If a general malfunction of the LifeStyle unit occurs, a half-second alarm sounds
every half second, and the red ALARM light turns on. When this occurs, oxygen
concentration is outside specifications. The patient should change to another sourceof oxygen and contact the Equipment Provider.

2.4

Oxygen Concentration Test and Specification
To ensure that LifeStyle’s oxygen output is within specification, you must perform a testof oxygen concentration. Test the unit upon delivery to a patient and at periodic intervals. Equipment Providers, based upon their own expertise and documentation, may
establish and implement their own protocol to check oxygen concentration. The interval
established may be longer or shorter than 90 days, AirSep’s default time period forproviders who do not choose to establish their own protocol.

1.

Connect LifeStyle to AC electrical power using only the AirSep-supplied AC
adapter (AirSep Part No.PW005) to a 100-250 volt, 50/60 Hz outlet.
(Note: Do not use the battery [AirSep Part No. BT007-1] to operate theunit when measuring oxygen concentration.)

2.

Turn on LifeStyle, and adjust the flow selector to any of 1 through 5 lpm
settings.

3.

Remove the gross particle filter.

4.

Activate the test mode by depressing the switch located in the hole labeled“T,” next to the hour meter. The unit should produce a pulse dose of oxygen
equivalent to a continuous flow of 4 lpm at a rate of 18 breaths-per-minute
while in the test mode.

5.

When the unit supplies a pulse dose of oxygen, the green PULSE light is on,
and you will be able to hear the pulse of oxygen. Verify that the pulse light isturning on and that you can hear the pulse dose.

6.

Allow LifeStyle to run for five minutes before recording the oxygen concentration. Connect a calibrated Teledyne TED or equivalent to the oxygen
outlet on the unit. Turn on the analyzer, and record the concentration. The
concentration specification in test mode is 90% ±3%.

2-4

3.0

Patient Instructions
3.1

General Instructions
It is important that patients thoroughly understand how to operate AirSep’s LifeStyle
unit. This enables proper treatment as prescribed by a qualified, licensed physician.
You must explain that the purpose of this therapy is to alleviate symptoms. If patients
experience any discomfort or if the unit alarms, they must notify their Equipment Provider and/or physician immediately.
You, as the Equipment Provider, are responsible to ensure that each patient receives the
LifeStyle Patient Manual. Explain each step in the operation of the unit to the patientwith reference to the patient manual.

3.2

Routine Maintenance by the Patient
To ensure accurate output and efficient operation of the unit, the patient must perform
two simple, routine maintenance tasks:•

Clean the air intake / gross particle filter•

Check the number of hours displayed on the hour meter.

3.2.1 Cleaning the Air Intake / Gross Particle Filter

NOTE
The patient must clean this filter weekly,
as described below. The filter may require
daily cleaning if the LifeStyle unit operates
in a harsh environment; for example, ahouse heated by wood, kerosene, or oil, or
one with excessive cigarette smoke.

1.

Lift the corner of the LifeStyle unit to expose the air intake / grossparticle filter.

3-1

2.

Remove the filter from the unit and wash it in a solution of warm waterand soap.

3.

Rinse the filter thoroughly, and remove excess water with a soft,absorbent towel. Ensure that the filter is dry before replacing it.

4.

Replace the dry filter.

3.2.2 Checking the Hour Meter
The compressor, product filter, and felt filter must be replaced after 3,000
hours of unit use. Instruct the patient to take note of the reading on the hourmeter, located beneath the air intake gross particle filter, whenever cleaning
the air intake / gross particle filter. When the reading approaches 3,000
hours, instruct the patient that it is important to notify the Equipment Provider.

3.3

Usage at Night or with Extended Tubing
Instruct patients that AirSep recommends that LifeStyle should be used while sleeping
or with extended tubing (up to 25 ft, 7.5 m) only under the direction of their clinician.

3-2

4.0

Equipment Provider Maintenance
4.1

Routine Maintenance
The LifeStyle unit includes two filters and a compressor that require scheduled maintenance or replacement.
To ensure that the output of oxygen is within specification, you must perform a test of
oxygen concentration. (Refer to Section 2.4, “Oxygen Concentration Test and Specification,” of this manual.) Test the unit upon delivery to a patient and at periodic intervals. Equipment Providers, based on their expertise and documentation, may establish
and implement their own protocol to check oxygen concentration The interval established may be longer or shorter than 90 days, AirSep’s default time period for providers
who do not choose to establish their own protocol.

4.1.1 Air Intake / Gross Particle Filter
The external air intake gross particle filter is located under the panel coveringthe LifeStyle controls. You can easily remove it by hand. Instruct the patient
to clean this filter weekly. (Refer to Section 3.2.1, “Cleaning the Air Intake /Gross Particle Filter,” of this manual.)

NOTE
The filter may require more frequent cleaning
if the LifeStyle unit operates in a harshenvironment — a house heated by wood,kerosene, or oil, or one with excessive cooking
or cigarette smoke.

4.1.2 Product Filter Replacement
The product filter must be replaced after every 3,000 hours of use.
1.

Set the unit’s ON/OFF switch to the OFF position, and disconnect
the power cord.

2.

Remove the bottom housing, and locate the product filter. (Refer to
Section 5.2.1 of this manual.)

3.

Disconnect the tubing from the inlet and outlet of the product filter.

4.

Connect the tubing to the new filter, and secure with tie wraps.

5.

Replace the bottom housing.

4-1

4.1.3 Compressor Replacement
As the Equipment Provider, you are responsible for monitoring the life of thecompressor, which must be replaced after 3,000 hours of use.
Refer to Section 5.3 of this manual for compressor replacement procedures.

4.1.4 Recording MaintenanceAs the Equipment Provider, you should record all routine maintenance andrepairs performed on the LifeStyle unit, including hours and dates of service.

4.2

Infection Control
With growing concern about possible cross infection from home oxygen equipment fromone home care patient to another, a clarification on this topic is necessary. The organisms of most concern are M. Tuberculosis, HIV, and Viral Hepatitis. These are potentially pathogenic.
Tuberculosis can survive outside the human body, but its mode of transmission is bydroplet nuclei. When infected individuals cough, they release droplet nuclei into the air,and these carry the Tuberculosis organism. These droplet nuclei may be breathed in byanother person, but prolonged exposure to the infected person is usually necessary for
infection to occur.
HIV and Viral Hepatitis are both viruses that are not living cells themselves but can
duplicate when in a living “host” cell. Both of these organisms are usually passed on by
person-to-person contact, and both need to be in the human body to survive. Onceoutside the body, viruses do not survive.

4.3

Cleaning and Preparing for New Patient Use
When you remove LifeStyle from a patient’s home, always detach and dispose of theused nasal cannula. Clean the exterior of LifeStyle with a mild household cleaner
applied with a damp cloth or sponge. Pay special attention to the oxygen outlet for thecannula connection to make sure it remains free of dust, water, and particles. Be careful
not to get any liquid into the interior of the unit.
Next, clean the exterior with a common chemical disinfectant before any other patientuses the unit. Do not use liquid directly on the LifeStyle unit to clean it.

4-2

A list of undesirable chemical agents includes, but is not limited to, the following,according to the plastic manufacturer: alcohol and alcohol-based products (isopropyl
alcohol), concentrated chlorine-based products (ethylene chloride), and oil-based
products (Pine-Sol, Lestoil). These are not to be used to clean the plastic housing of
LifeStyle. These chemicals and the cleaners that contain them will crystallize the plastic
and break down the plastic’s molecular properties.
Clean the air intake / gross particle filter with warm soapy water between each patient’s
use. The patients should be instructed to clean this filter at least once per week, depending on the environment, during normal operation.
Change the felt filter after 3,000 hours of use.
Change the product filter after 3,000 hours of use. It is not necessary to change thisfilter between patients even if the previous patient had a communicable disease or
infection.
Allow LifeStyle to air dry, and then retest oxygen concentration before you return theunit to your inventory.

4-3

5.0

Service
5.1

Components
The design of AirSep’s Lifestyle Portable Oxygen Concentrator allows for easy accessand removal of most components. This allows you to perform scheduled maintenanceand to repair and replace parts with minimal time and effort.

CAUTIONFor your safety, be sure to disconnect the
LifeStyle AC or DC adapter from the power
outlet and remove the battery before you
service the concentrator.

NOTE
Record all scheduled maintenance. (Refer
to Section 4.0 of this manual.)

5.2

Housing Removal
5.2.1 Removing Bottom Housing
1.

Place the unit on its top, upside-down, and locate the four rubber
feet on the bottom housing.

2.

Insert a Phillips-head screwdriver through each hole in the feet, andloosen the screw at the bottom of each hole. Repeat this for all four
feet.

3.

Unscrew the two screws in the center of the bottom housing, andremove the screws.

4.

Lift off the bottom housing.

5.2.2 Removing Top Housing
1.

Using a pair of pliers, gently pull the selector knob off.

2.

Turn the unit over so that it rests on the top housing.

5-1

5.3

3.

Use a Phillips-head screwdriver to remove the two screws on thecenter housing near the handle and the screw in the hole in thecenter housing near the muffler.

4.

Turn the unit over, and gently remove the top housing.

Compressor Replacement
The compressor is the “pump” within the oxygen concentrator that pushes room air into
the bottom of the sieve beds. This allows oxygen to flow out the top.
The LifeStyle compressor uses a brush-type DC motor. Over time, the brushes wear,
and the compressor does not operate properly. The expected life of the compressor is
3,000 hours.

5.3.1 Compressor Assembly Removal
To remove the compressor assembly for exchange, follow the steps
listed below:
1. Disconnect the LifeStyle unit from the AC or DC outlet, and remove thebattery.
2. Remove the top and bottom housing. (Refer to Section 5.2.)
3. Disconnect the suction tube.
4. Disconnect the orange 2-pin connector with the red and black wires
from the circuit board.
5. Remove the heat exchanger and clamp.
6. Remove the 3 hex nuts and split washers that hold the compressor to its
mounts.
7. Lift the compressor assembly, and stand up in the well of chassis.
8. Carefully disconnect the red compressor wires from the temperatureswitch.
9. Remove compressor assembly. Inspect the vibration mounts for damage.
Replace if necessary.
10. Transfer the brackets to the new compressor, leaving the screws loose.

5-2

5.3.2 Compressor Assembly Installation
To install the new compressor assembly, perform the compressor removal
procedure in reverse order.

NOTE
All washers (6) must be in their proper place before
tightening the mounting nuts. Once the compressor is in place,adjust it left and right to ensure that it is in a level position
before tightening the clamps.

5-3

6.0 Troubleshooting
6.1

General Troubleshooting
Before reviewing the Troubleshooting Chart, the following steps may be useful to isolate
any malfunctions:
1. Make sure all filters are clean.
2. Set the unit’s ON/OFF switch to the ON position. If the unit does not turn on,refer to the Troubleshooting Chart that follows.
3. Activate the test mode by inserting a paper clip through the hole marked “T” to theright of the hour meter.
4. Make sure that the unit is leak-free by testing all tubing connections and fittings withleak-testing solution. Protect the circuit board from solution, and start the leak testat the heat exchanger, following the air flow of the unit to the oxygen outlet. Repair
all leaks by tightening connections and fittings.
5. After three minutes, measure the oxygen concentration.
6. Review the Troubleshooting Chart that follows to isolate and repair any othermalfunctions.

6-1

6.2 Troubleshooting Chart
ProblemPossible Cause

Unit does not turn on,
and AC adapter isproperly connected toa wall outlet.

Wall switch power is off.

Turn on wall switch power.

Defective AC adapter.

Replace AC adapter.

AC adapter is not plugged into
LifeStyle.

Check connection from theAC adapter to LifeStyle.

General malfunction
alarm is activated.

Compressor failure.

Replace compressor.

Battery lasts lessthan 50 minutes.

Battery not fully charged.

Ensure that battery
charges for at least 2 to
2 ½ hours.

Battery failure.

Replace battery.

Oxygen concentration is
below specifications intest mode.

Compressor has reached its
expected life (~3,000 hrs).

Replace compressor.

Unit does not deliver
oxygen.

Cannula kinked or notconnected.

Check cannula and thecannula connection.

LifeStyle is not turned on.

Turn LifeStyle on to theprescribed setting.

Battery is dead.

Replace battery.

6-2

Solution

Appendix



Exploded Drawings
Figure 1: LifeStyle — Control Panel

A-1

Figure 2: LifeStyle — Internal, Top View

A-2

Figure 3: LifeStyle — Internal, Bottom View

A-3

Ref No.
12
3
45Part No.
BT007-1
CA186-1
CA188-1
CA199-1
CA211-1

Description
Battery, Assy, 3,000 mAh
Cabinet, Housing, Lower, Light Gray
Cabinet, Handle Cover, Dark Gray
Cabinet, Housing, Cover, Assy
Cabinet, Top Housing, Assy (logo, serial #,label, etc.)

Ref No.
67
8
9Part No.
FI109-1
FI110-1
MI187-1
MI195-1

Description
Filter, Foam, 15PPI, 2.35 x 2.35 x 1/2”
Filter, Felt, #10, Intake
Misc Knob, Selector Switch
Bumper, Rubber

Figure A.1: LifeStyle — Control Panel

A-1

Ref No.
1
2
3
4

Part No.
CB064-1
CO207-1
CL028-1
FN016-1

Description
Circuit Board, Assy, Tubing
Compressor, Assy
Clamp, AL, 3/16” OD Tube
Fan, Assy (12VDC), Brushless

Ref No.
5
6
7
8Part No.
HX011-2
MO016-2
SW002-1
VA319-1

Description
Heat Exchanger
Mount, Sandwich, Male-Female
Switch, Temperature, 145F NC
Valve, Assy, 12VDC

Figure A.2: LifeStyle — Internal, Top View

A-2

Figure A.3: LifeStyle — Internal, Bottom View

A-3