Speciﬁcations

OPTIONS
LE-200PS

Model
Interface

LE-150PS

Standard: RS-232C /RS-422 /RS-485

Optional: TTL (Model: “OP-5M”)

DSUB 25pin female connector (M2.6)

Measurement Connecter

ASYNC, ASYNC-PPP, Character
SYNC: SYNC/BSC, Bit SYNC: HDLC/SDLC/X.25

ASYNC, ASYNC-PPPBaud Rate

50bps~1Mbps (arbitrary)

50bps~500Kbps (arbitrary)

SYNC Clock

ST1 *1, ST2 *1, RT(RXC),
AR(clock extracted from the edge of SD/RD)-

Protocol2G byte SD Card
2G byte SD card.
*Same as the card packed with
LE-200PS/LE-150PS.

*The photo is for illustrative purposes only.

ASCII, EBCDIC, JIS7, JIS8, Baudot, Transcode, IPARS, EBCD, EBCDIK, HEX

Bit transmission order,Polarity

Order: LSB first/ MSB first (switchable), Polarity: Normal/ Invert (switchable)

Error Check

Parity (ODD, EVEN, MARK, SPACE), Framing, Break, Abort (LE-200PS only), Short frame (LE-200PS only), BCC (LRC,
CRC-6, CRC-12, CRC-16, CRC-ITU-T, FCS-16 (LE-200PS only) ), FCS-32 (LE-200PS only). BCC permeation mode

MemoryPC: Max. 8G byte on the HDD; PC-less: Capacity of the SD card
(specify the file size at 128K / 1M /2M / 4M / 8M byte)

Recording Type

Ring Buffer (continuous) mode, Fixed Buffer (full stop) mode

Mode

Remote mode (with PC); Data Logger mode (without PC)

Power Plug Cable

Measurement start/stop

Control from PC, Start/Stop switch, Auto-Power run, Specify date and time.

Idle TimeOFF (no-record); Resolution: 100ms, 10ms, 1ms; Max. 999. 9s

SIH-2PG

Time Stamp

OFF (no record); Date time stamp: “Day/Hr/Min”, “Hr/Min/Sec” or “Min/Sec/10ms”
Record signals (RS(RTS), CS(CTS), ER(DTR), DR(DSR), CD(DCD), CI(RI), EXIN) withtransmission/reception data. Waveform display (RS-232C only)
Record only frames of the specified address.
(HDLC/SDLC/X.25 only)

Address Filter
Protocol translation display on PC

PPP

Condition

Communication error, data string up to 8 characters (don’t care and bit mask available), idle time more than
the specified duration, matched timer/counter value, logic status of interface signal line, external signal.

ActionStop measurement (offset can be set), validate/invalidate trigger condition, control timer/counter, send
specified data string, send external signal, turn on/ off the light of user-defined LED

Retrieval function on PC

Communication error, data string up to 8 characters (don’t care and bit mask available), idle time morethan the specified duration, time stamp (don’t care available), trigger-matched data.

Trigger

Simulation

Transmit data registered in 16 data tables (16K byte) with one press of a key.
DTE/DCE mode selectable. Pre-set the timing of line/data. Insert parity error.

Conversion

Convert data into Text or CSV format and save.

LED

4 of two-color LED: Power/ Error, Test/Record, SD/RD, User defined U1/U2

SwitchOne: Run / STOP

External Trigger

Input: 1, Output: 2. 2.54mm pin header connector
2G byte2 – 8G byte *2

USB 2.0 Port

Mini-B connector High speed supported

Power *3

USB bus power or external DC(DC8 - 26.4V),
AC adapter: “3A-183WP09" (center +)Power consumption: Max. 1.2W, 0.1W/DC12V when powering off

Work time during Power failure

1 sec

Ambient Temperature, Humidity

In operation:-10~+55˚C In storage: -20~+60˚C, 5 - 85%RH (No condensation)

StandardCE(class A), EMC(EN61326-1:2006)

Dimensions, weight86(W)×130(D)×30(H) mm, approx. 220g
OS: Windows® XP/Vista/7

PC environment

3A-183WP09
Input: AC100〜240V, 50/60Hz
Output: DC9V, 2A
Plug: Center+, Outside diameter:
5.5mm, Inside diameter: 2.1mm

A PC-connectable communication protocol analyzer, which can also be
used as a stand-alone communication data logger that supports SD cards

DC Plug (Outside diameter:
5.5mm, Inside diameter:
2.1mm)<–>Y terminal: 1.8m
Supply external DC power tothe DC jack of the analyzer.
Cable clamp is included.

-

SDLC (modulo8/128),
ITU-T X. 25(modulo8/128), LAPD, PPP

SD CardLE-200PS
Communication Protocol Analyzer LE-150PS
RS-232C / RS-422 / RS-485

Wide Input AC Adapter

Data CodeLine Status Measurement®

SD-2GXDSUB 9pin
Monitor Cable

LE-259M1
Branch cable for monitoring
RS-232C over general
DSUB 9pin, such as with thePC.

Terminal Block Adapter

LE-5TBAn adapter to extract the
RS-422/485 signals on the
DSBU 25pin connector to theterminal block.
* Sync clock signal
measured by LE-200PS
cannot be captured.

TTL Monitor Probe Pod

OP-5M
A probe pod for monitoring
TTL-Level communicationlines at 2.5V/3.3V/5V,

Multi-Protocol Supported

LE-200PS* Cannot be used for
simulation function.

PC: PC/AT compatible*1: It is available only for measuring RS-232C. *2: 2GB SD card is supported. LINEEYE plans to upgrade it to 8GB SDHC card in the future.
*3: AC adapter is sold separately. In the Remote mode (with PC), it runs by the USB bus power. In the Logger mode (PC-less), you need tohave the optional AC adapter (3A-183WP09) or use the optional Power Plug Cable (SIH-2PG) and external DC power.

Baud Rate: Max. 1Mbps

DIN Rail Mounting
Plate for LE-series.

Async, Sync, BSC,
SDLC, HDLC, X.25, PPP

LE-DIN13

Standard Set

To mount the LE-150PS /LE-200PS on the 35mm DIN
rail.

PC-connectable Protocol Analyzer…x 1
DSUB 25pin monitor cable (LE-25M1)…x 1
Mini USB cable (SI-US218)…x 1
External signal I/O cable (LE-4TG)…x 1
2G byte SD card (SD-2GX)…x 1
PC software CD…x 1
Instruction Manual…x 1
Warranty…x 1
Read the instruction manual provided with the product before use and use theas explained in that manual. Using the product in ways not guaranteed in the
SAFETY productmanual, connecting it to systems outside of the specified ranges and remodeling can
all cause trouble and damage. LINEEYE CO., LTD. will assume no responsibility
WARNING whatsoever for trouble or damage arising because of unauthorized ways of use.

•All brand names and product names mentioned in this catalog are trademarks or registered trademarks of theirrespective companies. •Specifications and designs of products listed in this catalog are as of February 2012, andare subject to change without notice for improvement. •Colors of actual products may differ slightly from that
listed due to printing condition. •This catalog may not be reprinted or duplicated, in part or in whole.
©2012 by LINEEYE CO., LTD.

LINEEYE CO., LTD.

Async Communication Supported

Head Office/Sales Office
Marufuku Bldg, 5F, 39-1 Karahashi Nishihiragaki-cho, Minami-ku, Kyoto, 601-8468
PHONE: 81-75-693-0161 FAX:81-75-693-0163

LE-150PS

URL http://www.lineeye.com
E-mail : info@lineeye.co.jp

Baud Rate: Max. 500Kbps

Async, PPP* LINEEYE CO., LTD. is a venture company founded by electronicequipment development members of the former Sekisui ChemicalCo., Ltd. with investment from the Sekisui Venture Fund.

QR-00337
ER-00094

L-12202E/LE G

Printed in Japan

RS-232C / RS-422 / RS-485
Communication Protocol Analyzer

Supports RS-232C/RS-422/RS-485 (Standard Feature)
The device comes standard with RS-232C/RS-422/RS-485 interfaces,
which are used widely in medical equipment and electronic products.
With an optional TTL probe pod (OP-5M), it can monitor TTL-levelcommunication at 2.5V/ 3.3V/ 5V. LE-200PS can measure Sync
communication, which has transmission/ reception clocks.

Compact Size, Simple Operation
A lightweight communication protocol analyzer has a Data Logger function to record
data in the SD card for long hours.

External Signal
Terminal
SD CardSlot

RUN / STOP
SwitchDC Jack

USB
Connector

Two-color
LED

M3 Screw Holes *1

Measurement Connector*1: For attaching the optional DIN rail plate. Pitch: 70mm. Depth: 3.5mm (max).

Pin Assignment of Connector (DSUB 25 pin)
Pin

[ Terminal Block Adapter (LE-5TB) ]Signal1

FG

2

RS-232C SD

3

RS-232C RD

4

RS-232C RS

5

RS-232C CS

6

RS-232C DR

7

GND *2

8

RS-232C CD

9+5VDC *1

10

RS-422 RXDB+ *2

11

RS-422 RXDA- *2

12

RS-422 RXCB+ *3

13

RS-422 RXCA- *3

15

RS-232C ST2 *3

17

RS-232C RT *3

PLC

PC

Developing Device

SD

Robot

RD

SD
RD

Baud Rate[ TTL Monitor Probe Pod (OP-5M) ]*2

18

RS-422/485 TXDB+/TR+

19

RS-422/485 TXDA-/TR- *2

20

RS-232C ER

22

RS-232C CI

24

RS-232C ST1 *3*1: A terminal-supplied power from analyzer. ON/OFF
switchable.
*2: LE-5TB is useful for connecting the signals.
*3: LE-200PS only.
*4: Signal pins not defined here are not connected.

Measures at Arbitrary Speed.
Monitor data at any speed by setting the baud rate of any four digits.
The high-precision timer makes it possible to record idle time and timestamps along with data, and is not related to the performance of the PC.
[ Configuration Setting ]Remote Mode

Logger Mode
Analysis PC(PC connected)

Measured data is saved as log files of the specified file size in the HDD of
the PC (Remote mode) or in the SD card of the analyzer (Logger mode).
It automatically records data until reaching the specified number of files,
and then deletes the oldest[ Record Control Setting ]file to record the new file. Also,
it can stop measurement onreaching the specified numberof files. It is usefulfordetecting any hindrance inthe line.

[ Recording Time ]Two Types of Operations Based on Usage Situation.
This device can be used as a PC-connectable Protocol Analyzer for use in the lab (Remote mode), and also as a PC-less Data Logger for use at
on-site tests to record data for long hours (Logger mode).

Long Hour Recording

Record in the SD card

Capacity: 1.8G byte

9600bps

Approx. 100 Hours

230.4Kbps

Approx. 4 Hours(e.g. 4M byte×450 files)*: In the case of full-duplex transmission of 1K byte data at 1ms intervals.

Seamless access to communication log files
Communication log files can be viewed
in detail on a PC. It offers seamless
operation that handles a single
measurement log file even when all
files are read collectively. The
measured data can be converted into
text or CSV format to use in a word
processor and/or spreadsheet software.

[ Text Conversion Setting ]Easy-to-Operate Simulation Function
It incorporates an easy-to-use simulation function that makes it possible
to transmit preset transmission data (16 types of data), or fixed data such
as FOX messages, at the flip of a key while checking reception data.

(PC less)[ Example of Simulation ]It is useful in the situations of:Connect to PC
via USB

Record in the HDD of PC[ Data display example ]*There are space limitations.
*PC usage is not allowed.
*Dusty places (PC cannot be used).
*Need to record data for more than one month.
*Cannot operate the analyzer well

Schedule Measurement by Inner RTC
Real Time Clock (RTC) backed up by
the battery of the analyzer makes it
possible to specify the starting andending times of the measurement. Afterthe measurement, it turns off the power
automatically and saves on power
consumption.

Communication errors can be detected with high reliability.

[ Auto RUN/STOP setting ]Protects important data from corruption due to power failure
A newly developed instant power failure prevention circuit protects important
communication log files stored in the SD cards, by protecting the SD cards
from being corrupted even if power fails while recording data to the SD card.

Small and robust housing suitable for severe field environments
The palm-sized robust unit can be used between –10 to+55˚C. It operates not only on USB bus power, but also on
external DC power of 8 to 26.4V. The consumption current
is as low as 100mA at DC12V input. The SD card slot andthe USB connector are equipped with a dust-proof coverthat allows 2-way operation. It can be fixed easily to theequipment to be examined or built into an inspection line,
since it is compatible with 35mm DIN rails.

[ Onto 35mm DIN Rail ][ Data display example (Control line information is displayed.) ][ Dust-proof Covers ]It can judge parity error, framing error, BCC error of various block check
codes, and short SDLC/HDLC frame (LE-200PS only). It can find thecommunication sequence in the event of an error, by setting anapplication-level error notification character string to the character string
agreement criteria of the triggering function. It can notify an error toexternal devices and alert a communication error in the logger mode(with a panel LED lighting), by specifying an external trigger signal
output and user-defined LED lighting as a triggering action.
[ Trigger Setting ][ Simulation Data Tables ]RS-232C / RS-422 / RS-485
Communication Protocol Analyzer

Supports RS-232C/RS-422/RS-485 (Standard Feature)
The device comes standard with RS-232C/RS-422/RS-485 interfaces,
which are used widely in medical equipment and electronic products.
With an optional TTL probe pod (OP-5M), it can monitor TTL-levelcommunication at 2.5V/ 3.3V/ 5V. LE-200PS can measure Sync
communication, which has transmission/ reception clocks.

Compact Size, Simple Operation
A lightweight communication protocol analyzer has a Data Logger function to record
data in the SD card for long hours.

External Signal
Terminal
SD CardSlot

RUN / STOP
SwitchDC Jack

USB
Connector

Two-color
LED

M3 Screw Holes *1

Measurement Connector*1: For attaching the optional DIN rail plate. Pitch: 70mm. Depth: 3.5mm (max).

Pin Assignment of Connector (DSUB 25 pin)
Pin

[ Terminal Block Adapter (LE-5TB) ]Signal1

FG

2

RS-232C SD

3

RS-232C RD

4

RS-232C RS

5

RS-232C CS

6

RS-232C DR

7

GND *2

8

RS-232C CD

9+5VDC *1

10

RS-422 RXDB+ *2

11

RS-422 RXDA- *2

12

RS-422 RXCB+ *3

13

RS-422 RXCA- *3

15

RS-232C ST2 *3

17

RS-232C RT *3

PLC

PC

Developing Device

SD

Robot

RD

SD
RD

Baud Rate[ TTL Monitor Probe Pod (OP-5M) ]*2

18

RS-422/485 TXDB+/TR+

19

RS-422/485 TXDA-/TR- *2

20

RS-232C ER

22

RS-232C CI

24

RS-232C ST1 *3*1: A terminal-supplied power from analyzer. ON/OFF
switchable.
*2: LE-5TB is useful for connecting the signals.
*3: LE-200PS only.
*4: Signal pins not defined here are not connected.

Measures at Arbitrary Speed.
Monitor data at any speed by setting the baud rate of any four digits.
The high-precision timer makes it possible to record idle time and timestamps along with data, and is not related to the performance of the PC.
[ Configuration Setting ]Remote Mode

Logger Mode
Analysis PC(PC connected)

Measured data is saved as log files of the specified file size in the HDD of
the PC (Remote mode) or in the SD card of the analyzer (Logger mode).
It automatically records data until reaching the specified number of files,
and then deletes the oldest[ Record Control Setting ]file to record the new file. Also,
it can stop measurement onreaching the specified numberof files. It is usefulfordetecting any hindrance inthe line.

[ Recording Time ]Two Types of Operations Based on Usage Situation.
This device can be used as a PC-connectable Protocol Analyzer for use in the lab (Remote mode), and also as a PC-less Data Logger for use at
on-site tests to record data for long hours (Logger mode).

Long Hour Recording

Record in the SD card

Capacity: 1.8G byte

9600bps

Approx. 100 Hours

230.4Kbps

Approx. 4 Hours(e.g. 4M byte×450 files)*: In the case of full-duplex transmission of 1K byte data at 1ms intervals.

Seamless access to communication log files
Communication log files can be viewed
in detail on a PC. It offers seamless
operation that handles a single
measurement log file even when all
files are read collectively. The
measured data can be converted into
text or CSV format to use in a word
processor and/or spreadsheet software.

[ Text Conversion Setting ]Easy-to-Operate Simulation Function
It incorporates an easy-to-use simulation function that makes it possible
to transmit preset transmission data (16 types of data), or fixed data such
as FOX messages, at the flip of a key while checking reception data.

(PC less)[ Example of Simulation ]It is useful in the situations of:Connect to PC
via USB

Record in the HDD of PC[ Data display example ]*There are space limitations.
*PC usage is not allowed.
*Dusty places (PC cannot be used).
*Need to record data for more than one month.
*Cannot operate the analyzer well

Schedule Measurement by Inner RTC
Real Time Clock (RTC) backed up by
the battery of the analyzer makes it
possible to specify the starting andending times of the measurement. Afterthe measurement, it turns off the power
automatically and saves on power
consumption.

Communication errors can be detected with high reliability.

[ Auto RUN/STOP setting ]Protects important data from corruption due to power failure
A newly developed instant power failure prevention circuit protects important
communication log files stored in the SD cards, by protecting the SD cards
from being corrupted even if power fails while recording data to the SD card.

Small and robust housing suitable for severe field environments
The palm-sized robust unit can be used between –10 to+55˚C. It operates not only on USB bus power, but also on
external DC power of 8 to 26.4V. The consumption current
is as low as 100mA at DC12V input. The SD card slot andthe USB connector are equipped with a dust-proof coverthat allows 2-way operation. It can be fixed easily to theequipment to be examined or built into an inspection line,
since it is compatible with 35mm DIN rails.

[ Onto 35mm DIN Rail ][ Data display example (Control line information is displayed.) ][ Dust-proof Covers ]It can judge parity error, framing error, BCC error of various block check
codes, and short SDLC/HDLC frame (LE-200PS only). It can find thecommunication sequence in the event of an error, by setting anapplication-level error notification character string to the character string
agreement criteria of the triggering function. It can notify an error toexternal devices and alert a communication error in the logger mode(with a panel LED lighting), by specifying an external trigger signal
output and user-defined LED lighting as a triggering action.
[ Trigger Setting ][ Simulation Data Tables ]Speciﬁcations

OPTIONS
LE-200PS

Model
Interface

LE-150PS

Standard: RS-232C /RS-422 /RS-485

Optional: TTL (Model: “OP-5M”)

DSUB 25pin female connector (M2.6)

Measurement Connecter

ASYNC, ASYNC-PPP, Character
SYNC: SYNC/BSC, Bit SYNC: HDLC/SDLC/X.25

ASYNC, ASYNC-PPPBaud Rate

50bps~1Mbps (arbitrary)

50bps~500Kbps (arbitrary)

SYNC Clock

ST1 *1, ST2 *1, RT(RXC),
AR(clock extracted from the edge of SD/RD)-

Protocol2G byte SD Card
2G byte SD card.
*Same as the card packed with
LE-200PS/LE-150PS.

*The photo is for illustrative purposes only.

ASCII, EBCDIC, JIS7, JIS8, Baudot, Transcode, IPARS, EBCD, EBCDIK, HEX

Bit transmission order,Polarity

Order: LSB first/ MSB first (switchable), Polarity: Normal/ Invert (switchable)

Error Check

Parity (ODD, EVEN, MARK, SPACE), Framing, Break, Abort (LE-200PS only), Short frame (LE-200PS only), BCC (LRC,
CRC-6, CRC-12, CRC-16, CRC-ITU-T, FCS-16 (LE-200PS only) ), FCS-32 (LE-200PS only). BCC permeation mode

MemoryPC: Max. 8G byte on the HDD; PC-less: Capacity of the SD card
(specify the file size at 128K / 1M /2M / 4M / 8M byte)

Recording Type

Ring Buffer (continuous) mode, Fixed Buffer (full stop) mode

Mode

Remote mode (with PC); Data Logger mode (without PC)

Power Plug Cable

Measurement start/stop

Control from PC, Start/Stop switch, Auto-Power run, Specify date and time.

Idle TimeOFF (no-record); Resolution: 100ms, 10ms, 1ms; Max. 999. 9s

SIH-2PG

Time Stamp

OFF (no record); Date time stamp: “Day/Hr/Min”, “Hr/Min/Sec” or “Min/Sec/10ms”
Record signals (RS(RTS), CS(CTS), ER(DTR), DR(DSR), CD(DCD), CI(RI), EXIN) withtransmission/reception data. Waveform display (RS-232C only)
Record only frames of the specified address.
(HDLC/SDLC/X.25 only)

Address Filter
Protocol translation display on PC

PPP

Condition

Communication error, data string up to 8 characters (don’t care and bit mask available), idle time more than
the specified duration, matched timer/counter value, logic status of interface signal line, external signal.

ActionStop measurement (offset can be set), validate/invalidate trigger condition, control timer/counter, send
specified data string, send external signal, turn on/ off the light of user-defined LED

Retrieval function on PC

Communication error, data string up to 8 characters (don’t care and bit mask available), idle time morethan the specified duration, time stamp (don’t care available), trigger-matched data.

Trigger

Simulation

Transmit data registered in 16 data tables (16K byte) with one press of a key.
DTE/DCE mode selectable. Pre-set the timing of line/data. Insert parity error.

Conversion

Convert data into Text or CSV format and save.

LED

4 of two-color LED: Power/ Error, Test/Record, SD/RD, User defined U1/U2

SwitchOne: Run / STOP

External Trigger

Input: 1, Output: 2. 2.54mm pin header connector
2G byte2 – 8G byte *2

USB 2.0 Port

Mini-B connector High speed supported

Power *3

USB bus power or external DC(DC8 - 26.4V),
AC adapter: “3A-183WP09" (center +)Power consumption: Max. 1.2W, 0.1W/DC12V when powering off

Work time during Power failure

1 sec

Ambient Temperature, Humidity

In operation:-10~+55˚C In storage: -20~+60˚C, 5 - 85%RH (No condensation)

StandardCE(class A), EMC(EN61326-1:2006)

Dimensions, weight86(W)×130(D)×30(H) mm, approx. 220g
OS: Windows® XP/Vista/7

PC environment

3A-183WP09
Input: AC100〜240V, 50/60Hz
Output: DC9V, 2A
Plug: Center+, Outside diameter:
5.5mm, Inside diameter: 2.1mm

A PC-connectable communication protocol analyzer, which can also be
used as a stand-alone communication data logger that supports SD cards

DC Plug (Outside diameter:
5.5mm, Inside diameter:
2.1mm)<–>Y terminal: 1.8m
Supply external DC power tothe DC jack of the analyzer.
Cable clamp is included.

-

SDLC (modulo8/128),
ITU-T X. 25(modulo8/128), LAPD, PPP

SD CardLE-200PS
Communication Protocol Analyzer LE-150PS
RS-232C / RS-422 / RS-485

Wide Input AC Adapter

Data CodeLine Status Measurement®

SD-2GXDSUB 9pin
Monitor Cable

LE-259M1
Branch cable for monitoring
RS-232C over general
DSUB 9pin, such as with thePC.

Terminal Block Adapter

LE-5TBAn adapter to extract the
RS-422/485 signals on the
DSBU 25pin connector to theterminal block.
* Sync clock signal
measured by LE-200PS
cannot be captured.

TTL Monitor Probe Pod

OP-5M
A probe pod for monitoring
TTL-Level communicationlines at 2.5V/3.3V/5V,

Multi-Protocol Supported

LE-200PS* Cannot be used for
simulation function.

PC: PC/AT compatible*1: It is available only for measuring RS-232C. *2: 2GB SD card is supported. LINEEYE plans to upgrade it to 8GB SDHC card in the future.
*3: AC adapter is sold separately. In the Remote mode (with PC), it runs by the USB bus power. In the Logger mode (PC-less), you need tohave the optional AC adapter (3A-183WP09) or use the optional Power Plug Cable (SIH-2PG) and external DC power.

Baud Rate: Max. 1Mbps

DIN Rail Mounting
Plate for LE-series.

Async, Sync, BSC,
SDLC, HDLC, X.25, PPP

LE-DIN13

Standard Set

To mount the LE-150PS /LE-200PS on the 35mm DIN
rail.

PC-connectable Protocol Analyzer…x 1
DSUB 25pin monitor cable (LE-25M1)…x 1
Mini USB cable (SI-US218)…x 1
External signal I/O cable (LE-4TG)…x 1
2G byte SD card (SD-2GX)…x 1
PC software CD…x 1
Instruction Manual…x 1
Warranty…x 1
Read the instruction manual provided with the product before use and use theas explained in that manual. Using the product in ways not guaranteed in the
SAFETY productmanual, connecting it to systems outside of the specified ranges and remodeling can
all cause trouble and damage. LINEEYE CO., LTD. will assume no responsibility
WARNING whatsoever for trouble or damage arising because of unauthorized ways of use.

•All brand names and product names mentioned in this catalog are trademarks or registered trademarks of theirrespective companies. •Specifications and designs of products listed in this catalog are as of February 2012, andare subject to change without notice for improvement. •Colors of actual products may differ slightly from that
listed due to printing condition. •This catalog may not be reprinted or duplicated, in part or in whole.
©2012 by LINEEYE CO., LTD.

LINEEYE CO., LTD.

Async Communication Supported

Head Office/Sales Office
Marufuku Bldg, 5F, 39-1 Karahashi Nishihiragaki-cho, Minami-ku, Kyoto, 601-8468
PHONE: 81-75-693-0161 FAX:81-75-693-0163

LE-150PS

URL http://www.lineeye.com
E-mail : info@lineeye.co.jp

Baud Rate: Max. 500Kbps

Async, PPP* LINEEYE CO., LTD. is a venture company founded by electronicequipment development members of the former Sekisui ChemicalCo., Ltd. with investment from the Sekisui Venture Fund.

QR-00337
ER-00094

L-12202E/LE G

Printed in Japan