FN3002
Load Cell Tension and Compression
SPECIFICATIONS






FEATURES

Static and dynamic applications
Rod end available in option
Integrated Amplifier optional
IP65 optional

APPLICATIONS





Process control equipment
Cable stress tests benches
Shock absorbers tests benches
Structure fatigue tests
Laboratory and Research

SENSOR SOLUTIONS /// FN3002_en

Standard ranges 10kN to 2000kN or[2 klbf to 400 klbf]
Harsh industrial environment
Very high stability
Stainless steel or aluminum
IP protection available
High level output with integrated amplifier

The FN3002 is suitable for measuring cable tension and for use
in test benches and in many industrial applications. The FN3002
with its compact dimensions measures tension and compression
in standard ranges from 10 kN to 2 000 kN [2 klbf to 400 klbf].
A model with integrated amplifier for high-level output is alsoavailable.
With a long standing experience as a designer andmanufacturer of sensors, TE CONNECTIVITY often works withcustomers to design or customize sensors for specific uses andtesting environments.
On request, Instruction documents can be provided toease the selection and use of our sensors and provide helpful
tips.

05/2018

Page 1FN3002

Load Cell Tension and Compression

PERFORMANCE SPECIFICATIONS (typical values at temperature 23°C)
Ranges (FS) (N)

10k

20k

50k

100k

200k

500k

1000k

2000k

Ranges (lbf)

2.2k

4.5k

11.2k

22.5k

45.0k

112.40k224.81k449.6kMaterial

Aluminium

Stainless steel

Stiffness (N/m)

2.3E+08 3.4E+08 1.0E+09 1.5E+09 2.3E+09 3.7E+09 5.6E+09 8.9E+09

Stiffness (lbf/ft)

1.5E+07 2.3E+07 6.9E+07 1.0E+08 1.6E+08 2.5E+08 3.8E+08 6.1E+08

Standard electrical output

JAEGER 4B
VersionJAEGER 7B

Standard

A1A2Power supply

10Vdc

10Vdc to 30Vdc±12Vdc to ±18Vdc

Sensitivity (FSO)±20mV±2Vdc ±0.2V±5Vdc ±0.25V

Offset<±1 mV

2.5Vcc ±0.2V

0V ±0.25V

Input Impedance / Consumption

350 to 700 ohms

<30mA

Output Impedance

350 to 700 ohms

1 kohms max

Overrange Without Damage

1.5x FS

Overrange Without Destruction

3x FS

Combined Non-Linearity & Hysteresis< ±0.25% FS

Operating Temperature Range (OTR)-20°C à +80°C (-4 to 176°F)

Compensated Temperature Range (CTR)

0°C à +60°C (32 to 140°F)

Thermal Zero Shift in CTR< 0.5% FS/50°C

Thermal Sensitivity Shift in CTR< 1%/50°C

Insulation> 1000 Mohms

Protection Index

IP50Notes
1. Signal goes positive in tension with standard wiring configuration. Other signal output on request
2. Electrical Termination: Connector output including mate
3. Materials: Body in stainless steel or aluminium alloy depending on F.S ; aluminum cover
4. Output impedance < 100Ω on request
5. CE conformance according to EN 61010-1, EN 50081-1, EN 50082-1

SENSOR SOLUTIONS /// FN3002_en

05/2018

Page 2

FN3002

Load Cell Tension and Compression

DIMENSIONS & WIRING SCHEMATIC (IN METRIC AND IMPERIAL)
Wiring Schematic

Dimensions in mm [inch]
Ranges in N

10k

20k

50k

100k

200k

500k

1000k

2000k

A

70 [2.76]

75 [2.95]

80 [3.15]

85 [3.35]

105 [4.13]

140 [5.51]

160 [6.30]

B

40 [1.57]

50 [1.97]

55 [2.17]

65 [2.56]

85 [3.35]

115 [4.53]

150 [5.91]

C (Thread)

M20x1.5

M22x1.5

M35x1.5

M45x1.5

M64x4

M80x2

M110x2

D

25 [0.98]

25 [0.98]

32 [1.26]

40 [1.57]

70 [2.76]

100 [3.94]

115 [4.53]

E

22 [0.87]

30 [1.18]

36 [1.42]

48 [1.89]

66 [2.60]

85 [3.35]

120 [4.72]

F

30 [1.18]

40 [1.57]

45 [1.77]

55 [2.17]

75 [2.95]

95 [3.74]

130 [5.12]

SENSOR SOLUTIONS /// FN3002_en05/2018

Page 3

FN3002

Load Cell Tension and Compression

OPTIONSZ04

CTR -40°C to +90°C (OTR -40°C to +90°C)

Z35CTR +20°C to +120°C (OTR -20°C to +120°C)

Z36

CTR +20 to +150°C (OTR -20°C to +150°C) - Not available on A1 / A2 versions

ORDERING INFO

FN3002 - A1 - 10KN- /Z04
Options (Z04,...)
Range in Newton
Amplified version (none, A1 or A2)
Model

SUPPLIED ACCESSOIRESEFMX-4M : mating plug Jaeger 530-801-006 with clamp 530-841-006 for standard or Z04 (ranges 10kN to 200kN)
EFMX-4H : mating plug Jaeger 530-804-006 with clamp 530-844-006 for Z35 or Z36 option (ranges 10kN to 200kN)
EFMX-7M : mating plug Jaeger 530-272-006 with clamp 530-371-006 for standard and Z04 (ranges 500kN to 2MN)
EFMX-7H : mating plug Jaeger 530-604-006 with clamp 530-693-006 for Z35 or Z36 option (ranges 500kN to 2MN)

RECOMMENDED ACCESSORIESOPTERxxx: Rod Ends (Contact us)

TE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred to herein
might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard
Terms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuseof the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies

SENSOR SOLUTIONS /// FN3002_en

All Rights Reserved.

05/2018

Page 4