FN2570
Extra-flat Pedal Load Cell
SPECIFICATIONS

Range 200 to 2500 N (40 to 500 lbf)
Stainless Steel material
Cable Output
Very easy fixation

The FN2570 load cell has been developed for pedal force
testing. It attaches easily on all types of pedals and test rigs
for endurance and fatigue tests, especially where size is an
important factor.
The force is measured with high precisionindependent of the point of application of force. A strain
relief spring strengthens the cable output.

FEATURES

Compression onlyAccuracy regardless of forceapplication point
Extra flat and rugged design

A similar pedal load cells with a better accuracy
1%FS and integrated amplifier option exists as model
FN2114.

APPLICATIONSWith many years of experience as a designer andmanufacturer of sensors, Measurement Specialties, Inc.
often works with customers to design or customize sensors
for specific uses and testing environments.

On-board vehicle equipment
Test bench equipment
Laboratory and research

On request, Instruction documents can be provided
to ease the selection and use of our sensors and provide
helpful tips.

SENSOR SOLUTIONS /// FN2570_en06/2017

Page 1

FN2570
Extra-flat Pedal Load Cell

STANDARD RANGES
Ranges in N (FS)

200500

1000

1500

2000

2500

Ranges in lbf (FS)

40100200300400500

MaterialAluminum

Stainless steel

PERFORMANCE SPECIFICATIONS (typical values at temperature 23±3°C)
Parameters
Operating Temperature Range (OTR)-20 to 80° C [-4 to 176° F]

Compensated Temperature Range (CTR)

0 to 60º C [32 to 140° F]

Zero Shift in CTR<0.5% F.S. / 50º C [/100° F]

Sensitivity Shift in CTR<1% of reading / 50º C [/100° F]

Range (F.S.)

0-200 to 0-2500 N [0-40 to 0-500 lbf]

Over-Range
Without Damage

1.5 x F.S.

Without Destruction

3 x F.S.

Accuracy
Combined non-linearity & hysteresis

±2.5% F.S.

Electrical Characteristics
Model

FN2570

Supply Voltage

10Vdc

F.S. Output±2mV/V

Zero Offset±5% F.S.

Input Impedance/Consumption

700Ω

Output Impedance

700Ω

Insulation under 50Vdc≥100MΩ

Notes
1. Electrical Termination: 2m shielded cable length
2. Protection Index: IP50 3.
3. CE conformance according to EN 61010-1, EN 50081-1, EN 50082-1

SENSOR SOLUTIONS /// FN2570_en

06/2017

Page 2

FN2570
Extra-flat Pedal Load CellDIMENSIONS & WIRING SCHEMATIC (IN METRIC AND IMPERIAL)

FN2570

FN2570-6

Wiring Schematic

SENSOR SOLUTIONS /// FN2570_en

06/2017

Page 3

FN2570
Extra-flat Pedal Load CellOPTIONS
FN2570: Cable output with strait tube
FN2570-6: Cable output with spring
L00M : special cable length, replace “00” with total length in meters

ORDERING INFORMATION
FN2570-6-

1KN-

Model-Range/Unit-

FN2570
FN2570-6-200N
500N
1KN
1.5KN
2KN
2.5KN

-/L5M
Options/L00M

TE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred to herein
might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard
Terms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuseof the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies

SENSOR SOLUTIONS /// FN2570_en

All Rights Reserved.

06/2017

Page 4