AGILENTROTARY VANE PUMPS

2-3

Features and Benefits

4-5

Typical Applications

6-19

Pump Models

20

Accessories

21

HS Series Pumps Features and Benefits

22-25 HS Series Pump Models
26-27 MS 40+ Pump Features and Benefits
28-29 MS 40+ Pump Model

ROTARY VANE PUMPSAGILENT ROTARY VANE PUMPS FEATURES AND BENEFITS● Thanks to a very simple and highly reliable design, these field provenrotary vane pumps provide excellent vacuum performance. Agilent’s
quality and manufacturing standards ensure that the DS Rotary Vane
Pumps provide high pumping stability for light gases, low noise, minimal
oil backstreaming, and a long operating life.
● Agilent’s DS Rotary Vane Pumps conform with CE and RoHS
requirements, and all pumps are UL and CSA approved. Agilent’s worldclass technical support organization makes the DS Rotary Vane Pump
cost-effective and well suited for a wide range of applications.

The new
MS 40+
Mono Stage
Rotary Vane Pump

HS Series“smart”
pumps withgreen technology• High capacity pumping speed withthe smallest footprint• Proven inverter technology• Ideal for mass spectrometry andelectron microscopy• Lowest noise, highest throughput• In 2004 Varian, now Agilent,introduced the first rotary
vane pumps with truly “smart”
capabilities• Employ an innovative frequency
inverter technology to deliver
optimal and consistent
performance• Encompasse the worldwide rangeof voltage and frequency conditions• Enviromentally friendly thanks to
reduced power requirements andlow start up current

Dual Stage Pumps
DS Series: the two stage design allows:• Low 10-3 mbar operation• Low operating temperature• Minimal backstreaming at low pressure• Good pumping efficiency and gas ballast
in the low 10-2 mbar region

Worldwide Motors• Pump motors, available as 1-phase or
3-phase, are suitable for all voltages andfrequencies worldwide• Operational voltages are easily
selectable, allowing greater flexibility,
easy planning and inventory reduction

Anti-suckback Valve and Vent Device

Socket Type IEC320• This valve isolates the pump should itstop or be idle

• Permits use of standard power cable• Eliminates the need to open box andwire the motor• Prevents inadequate venting and oil
contamination of the vacuum system
when the pump is switched off, or incase of power failForced Oil Circulation

Gas Ballast Valve• The dedicated oil circulation gear pumpensures efficient and reliable lubricationof the pump from atmospheric pressure
throughout the entire vacuum operating
range• The opening of this valve injects dry air
into the second stage of the pump
• This action increases the temperatureof the module which facilitates theoutgassing and clean up of water vapor
or other condensable gases from the oil

Built-in Oil Shield

Alignment Pins on Pump Module• This feature minimizes the oil mist at thepump exhaust• The pump module components areassembled and positioned by built-inalignment pins

• Drastically reduces the oil consumption
over long periods of operation• Reduction of air pollution limiting theimpact on the environment
2

Rotary Vane
Pumps

A reliable line of pumps to cover the mostdemanding industrial and scientific applications• These speed up the assembling andmaintenance process, avoiding any error

Forced Air Ventilation
The cooling fan between the motor andthe pump:• Reduces the pump operating
temperature• Lowers the oil vapor partial pressure• Minimizes oil backstreaming and vacuum
system contamination
3

ROTARY VANE PUMPS

Rotary Vane
Pumps

AGILENT ROTARY VANE PUMPS TYPICAL APPLICATIONSICP-MS. Photo courtesy Agilent Scientific Instruments.

Freeze Drying equipment.

Analytical Instruments
and Mass Spectrometry
Rotary Vane Pumps are the most common primaryvacuum pumps used on GC-MS, LC-MS, ICP-MS, and
MALDI-TOF Instruments.
GC-MS typically uses our smallest pump, the DS42; thepump is needed to rough the system and back the highvacuum Turbo or Diffusion pump.
LC-MS and ICP-MS use a medium capacity pump on thesample injection/system interface, typically a DS402 or a
DS602, and a smaller pump to back the system Turbo pumps.
MALDI-TOF depending on system size, uses the DS102 to
the DS602 as roughing or interface pumps.
Nowadays high-end instruments, such as LC-MS and
ICP-MS, can take advantage of the benefits of HS 452,
HS 652 and MS 40+:
• Large pumping capacity• Consistent worldwide performance thanks to universal
voltage and frequency• Single phase• Low power requirements and start up current• Remote control and diagnostic• Adjustable performance, low noise• Green technology

Electron Microscopes

Industrial Vacuum Processes

Small pumps, typically the DS202 and the DS302,
are still used in competition with dry pumps to roughthe system and the high vacuum Turbo or Diffusion pump.
condensable vapor.

Vacuum coatings, Metallurgy Vacuum Furnaces, Lamps andTV Tube manufacture, Sterilizers, Freeze Dryers, Glove Boxes,
High Speed Centrifuges, and Flywheels (for energy storage)
represent the main Industrial applications for Dual Stage
Rotary Vane pumps. In the first four fields of application
the Rotary Pumps are used as roughing and backing pumps
for High Vacuum Turbo or Diffusion pumps, while in theothers the Rotary Pumps are typically the only vacuum pumps
on the system.

Leak Detection
Pumps in the DS102, DS202 and DS302 range are
typically used to back the Helium Mass Spectrometer Turboor Diffusion pump.
The roughing pump is usually bigger, typically up to theDS402/DS602, and can still be installed on the Leak Detector
itself, while bigger pumps can be used to pump down highthroughput Leak Detection systems.

4

Distillation apparatus. Photo courtesy University of Torino, Italy.

High Energy Physics and ResearchLaboratories
In these applications the Rotary Vane pumps are mainly usedin combination with Turbo pumps. The combination istypically used to rough and pump High Vacuum experimental
chambers or to start Ion Pumps in ultra high vacuum systems.

OLED-Lab. Photo courtesy PNNL.

5

DS 40M

DS 102

DS 202

DS 302

DS 402

DS 602HS 452HS 652

MS 40+

36 (1.27)
43 (2.58)

114 (4)95 (5.7)

192 (6.8)
160 (9.6)285 (10)237 (14.2)

410 (14.5)
342 (20.5)

605 (21.4)
504 (30.2)

456 (16.1)
456 (27.3)

672 (23.8)
672 (40.3)

828 (29.2)
828 (49.7)

1.27
1.8

3.5
5

5.8
8.3

8.2
11.6

12.3
17.4

17.6
25

13
22

19
32

23.5
40

Ultimate partial pressure* (mbar)

–

10-4

10-4

10-4

10-4

10-4

10-4

10-4

N/A

Ultimate total pressure* (mbar)

6.7 x 10-3

2 x 10-3

2 x 10-3

2 x 10-3

2 x 10-3

2 x 10-3

2 x 10-3

2 x 10-3

5 x 10-2

Ultimate total pressure with gas ballast* (mbar)–

2 x 10-2

2 x 10-2

2 x 10-2

1 x 10-2

1 x 10-2

1 x 10-2

1 x 10-2

N/A
No gas ballast port

Water vapor tolerance (mbar)

–

15

15

20

30

30

30

30

N/A
No gas ballast port

Water vapor capacity (g/h)

–

60

100

160

350

550

350

550

N/A
No gas ballast port

Oil capacity min/max (l)0.37 (max)

0.2/0.5

0.2/0.6

0.25/0.6

0.5/1

0.5/1

0.5/1

0.5/1

1

0.1
0.1

0.45
0.38

0.45
0.38

0.45
0.38

0.90
0.75

0.90
0.75

0.90
0.75

0.90
0.75

0.50

0.50

0.75

Free air displacement

60 Hz l/min (cfm)
50 Hz l/min (m3/h)

Pumping speed*

60 Hz (cfm)
50 Hz (m3/h)

Motor rating 1 ph

60 Hz (kW)50 Hz (kW)

Motor rating 3 ph

60 Hz (kW)50 Hz (kW)Nominal rotation speed

60 Hz (rpm)50 Hz (rpm)

3300
2600

1800
1500

1800
1500

1800
1500

1800
1500

1800
1500

2000

2000

1450

Weight kg (lbs)

9.3 (20.5)

22 (48)25 (55)25 (55)35 (77)35 (77)33 (73)33 (73)33 (73)

Inlet flange

16KF DN

25KF DN

25KF DN

25KF DN

25KF DN

25KF DN

25KF DN

25KF DN

25KF DN - 40KF DN

Exhaust flange

16KF DN

25KF DN

25KF DN

25KF DN

25KF DN

25KF DN

25KF DN

25KF DN

25KF DN*According to PNEUROP 6602

6

7

Rotary Vane
Pumps

ROTARY VANE PUMPSAGILENT ROTARY VANE PUMP MODELSROTARY VANE PUMPSAGILENT ROTARY VANE PUMP MODELS► Agilent DS 40M

DS 40M - Pumping Speed Curve

107 (4.21)==
120 (4.72)

RUBBERFEET

Ø5 (0.19)

120 (4.72)

86,5 (3.40)

7.8 (0.30)

288.8 (11.37)
INLET FLANGE 16KF DN

100 (3.93)

10-2

1

101

102

1.76572.0

1.17711.0
0.8

0.5885(m3/h) 0.6

0.2942

0.4

0.1
10 -3

23.5 (0.92)(cfm)

0.17650.2

29 (1.14)
OUTLET FLANGE 16 KF DN

0.1177

10 -2

10 -1

OIL INLET PLUG 3/4G

1

(mbar)

101

10 2

10 3

0.0588Pressure
50 Hz

Dimensions: millimeters (inches)Technical Specifications
Free air displacement
Pumping speed*
Ultimate partial pressure*
Ultimate total pressure*
Ultimate total pressure with gas ballast*
Noise level
Oil capacity max
Motor rating 1ph
Nominal rotation speed
WeightInlet flange
Exhaust flange(Torr)

10-1

Rotary Vane
Pumps

Pumping Speed

202.4 (7.96)

154.2 (6.07)

36.9 (1.45)

157.8 (6.21)

OIL DRAIN
PLUG

POWER
SWITCH

163.3 (6.43)

POWER INLET(EN 60320)

OIL LEVEL

3.0

10-3

60 Hz

Ordering Information
60 Hz: 36 l/min (1.27 cfm)
60 Hz: 1.27 cfm

60 Hz: 46 dB(A)60 Hz: 0.1 kW
60 Hz: 3300 rpm

50 Hz: 43 l/min (2.58 m3/h)
50 Hz: 1.8 m3/h–
6.7 x 10-3 mbar–
0.37 l9.3 kg (20.5 lbs)
16KF DN
16KF DN

Dual Stage Rotary Vane Pump		
DS 40M 100-120 Vac - 50/60Hz		DS 40M 200-240 Vac - 50/60Hz		

Part Number
X3703-64000
X3703-64001

Oil and Accessories		
Rotary vane fluid, AVF 20S type, 0.5 liter		DS 40M oil mist trap 3/4 G		
DS 40M oil mist trap cartdrige (pkg. of 2)		DS 40M maintenance kit		DS 40M oil drain extension		
NW 16 centering ring viton		NW 16 aluminum clamp		

50 Hz: 45 dB(A)50 Hz: 0.1 kW
50 Hz: 2600 rpm* According to PNEUROP 6602

DS 42 - Pump Speed Curve

8

9Part Number
X3703-64006
X3703-64003
X3703-64004
X3703-64005
X3703-64007
KC16AVKQ16AWPROTARY VANE PUMPSAGILENT ROTARY VANE PUMP MODELS► Agilent DS 102

DS 102 - Pumping Speed Curve(Torr)

30 (1.18)

35 (1.38)

3 (0.12)3 (0.12)35 (1.38)

212 (8.3)

212 (8.3)

104 (4.1) (2)104 (4.1)
132 (5.2)(2)245 (9.6) (1)245 (9.6)

132 (5.2)

10-2

10-1

1

101

50

165 (6.5)(1)105 (4.13) 105 (4.13)

430 (16.9) 430 (16.9)

Effective Pumping Speed 50 Hz

30 (1.18) 30 (1.18)

30 (1.18)

165 (6.5)

10-3

Rotary Vane
Pumps

100

10

10

5

(m3/h)

(cfm)
1
1

0.5

0.1
0.1

Alternative mounting holes are also available,with (1) = 226 (8.9) and (2) = 98 (3.8), with hole Ø = 7 (0.3)

10 -3

10 -2

10 -1

1

10 1

10 2

(mbar)
Pressure

Dimensions: millimeters (inches)Technical Specifications
Free air displacement
Pumping speed*
Ultimate partial pressure*
Ultimate total pressure*
Ultimate total pressure with gas ballast*
Water vapor tolerance
Water vapor capacity
Oil capacity min/max
Motor rating 1ph
Nominal rotation speed
WeightInlet flange
Exhaust flange

Ordering Information
60 Hz: 114 l/min (4 cfm)
60 Hz: 3.5 cfm

60 Hz: 0.55 kW
60 Hz: 1800 rpm

10-4 mbar
2 x 10-3 mbar
2 x 10-2 mbar
15 mbar
60 g/h
0.2/0.5 l

22 kg (48 lbs)
25KF DN
25KF DN

50 Hz: 95 l/min (5.7 m3/h)
50 Hz: 5 m3/h

Dual Stage Rotary Vane Pump		
DS 102 with 1 phase worldwide motor*		

Part Number
9499315

Oil and Accessories		
Rotary vane fluid, DS19 type, 1 liter		Rotary vane fluid, DS19 type, 1 liter (USA)*		
Rotary vane fluid, DS19 type, 1 gallon (USA)		
Oil mist eliminator		Oil mist replacement cartridge (pkg. of 2)		NW 25 oil exhaust filter		
NW 25 oil exhaust replacement cartridge		
Oil return kit		Oil drain extension		
European plug power cable 2 meters 1 ph		
USA plug power cable 2 meters 1 ph		
UK plug power cable 2 meters 1 ph		
NW 25 centering ring Viton		NW 25 aluminum clamp		

* 1 phase motors (100-120 / 200-230) V ±10%, 50/60 Hz.
All motors comply with CE and UL/CSA standards.

50 Hz: 0.45 kW
50 Hz: 1500 rpm

Minor Maintenance Kit		
Contains all the valves, O-rings and seals
to refurbish the pump to vacuum integrity		

Part Number

Major Maintenance Kit		
Includes all the items of the minor kit plus the vanes

Part Number
9499380

9499370* According to PNEUROP 6602

Part Number
9499390K7516301
K7516302
9499395
9499394
9499392
9499342
9499376
9499375
9499396
9499397
9499398
KC25AV
KQ25AWP* When these pumps are used in Leak Detectors applications, werecommend the use of Rotary Vane Fluid (Elite-Z mechanical),P/N 695409005 as it features a lower vapor pressure.

10

11

ROTARY VANE PUMPSAGILENT ROTARY VANE PUMP MODELS► Agilent DS 202

DS 202 - Pumping Speed Curve(Torr)
10-3

10-2

10-1

1

101

50

Rotary Vane
Pumps

100

3 (0.12)35 (1.38) 35 (1.38)

212 (8.3)

212 (8.3)

3 (0.12)30 (1.18) 30 (1.18) 30 (1.18) 30 (1.18)

104 (4.1) (2)104 (4.1) (2)245 (9.6) (1)245 (9.6) (1) 105 (4.13) 105 (4.13)

132 (5.2) 132 (5.2)

467 (18.4) 467 (18.4)

Effective Pumping Speed 50 Hz

205 (8.1) 205 (8.1)

10

10

5

(m3/h)

(cfm)
1
1

0.5

0.1
0.1

Alternative mounting holes are also available,with (1) = 226 (8.9) and (2) = 98 (3.8), with hole Ø = 7 (0.3)

10 -3

10 -2

10 -1

1

10 1

10 2

(mbar)
Pressure

Dimensions: millimeters (inches)Technical Specifications
Free air displacement
Pumping speed*
Ultimate partial pressure*
Ultimate total pressure*
Ultimate total pressure with gas ballast*
Water vapor tolerance
Water vapor capacity
Oil capacity min/max
Motor rating 1ph
Nominal rotation speed
WeightInlet flange
Exhaust flange

Ordering Information
60 Hz: 192 l/min (6.8 cfm)
60 Hz: 5.8 cfm

60 Hz: 0.55 kW
60 Hz: 1800 rpm

50 Hz: 160 l/min (9.6 m3/h)
50 Hz: 8.3 m3/h

10-4 mbar
2 x 10-3 mbar
2 x 10-2 mbar
15 mbar
100 g/h0.25/0.6 l

25 kg (55 lbs)
25KF DN
25KF DN

Dual Stage Rotary Vane Pump
DS 202 with 1 phase worldwide motor*

Part Number
9499320

Oil and Accessories		
Rotary vane fluid, DS19 type, 1 liter		Rotary vane fluid, DS19 type, 1 liter (USA)*		
Rotary vane fluid, DS19 type, 1 gallon (USA)		
Oil mist eliminator		Oil mist replacement cartridge (pkg. of 2)		NW 25 oil exhaust filter		
NW 25 oil exhaust replacement cartridge		
Oil return kit		Oil drain extension		
European plug power cable 2 meters 1 ph		
USA plug power cable 2 meters 1 ph		
UK plug power cable 2 meters 1 ph		
NW 25 centering ring Viton		NW 25 aluminum clamp		

* 1 phase motors (100-120 / 200-230) V ±10%, 50/60 Hz.
All motors comply with CE and UL/CSA standards.

50 Hz: 0.45 kW
50 Hz: 1500 rpm

Minor Maintenance Kit
Contains all the valves, O-rings and seals
to refurbish the pump to vacuum integrity

Part Number

Major Maintenance Kit
Includes all the items of the minor kit plus the vanes

Part Number
94993819499370* According to PNEUROP 6602

Part Number
9499390K7516301
K7516302
9499395
9499394
9499392
9499342
9499376
9499375
9499396
9499397
9499398
KC25AV
KQ25AWP* When these pumps are used in Leak Detectors applications, werecommend the use of Rotary Vane Fluid (Elite-Z mechanical),P/N 695409005 as it features a lower vapor pressure.

12

13

ROTARY VANE PUMPSAGILENT ROTARY VANE PUMP MODELS► Agilent DS 302

DS 302 - Pumping Speed Curve(Torr)
10-3

10-2

10-1

1

101

50

Rotary Vane
Pumps

100

3 (0.12)35 (1.38) 35 (1.38)

212 (8.3)

212 (8.3)

3 (0.12)30 (1.18) 30 (1.18) 30 (1.18) 30 (1.18)

104 (4.1) (2)104 (4.1) (2)245 (9.6) (1)245 (9.6) (1) 105 (4.13) 105 (4.13)

132 (5.2) 132 (5.2)

467 (18.4) 467 (18.4)

Effective Pumping Speed 50 Hz

205 (8.1) 205 (8.1)

10

10

5

(m3/h)

(cfm)
1
1

0.5

0.1
0.1

Alternative mounting holes are also available,with (1) = 226 (8.9) and (2) = 98 (3.8), with hole Ø = 7 (0.3)

10 -3

10 -2

10 -1

1

10 1

10 2

(mbar)
Pressure

Dimensions: millimeters (inches)Technical Specifications
Free air displacement
Pumping speed*
Ultimate partial pressure*
Ultimate total pressure*
Ultimate total pressure with gas ballast*
Water vapor tolerance
Water vapor capacity
Oil capacity min/max
Motor rating 1ph
Nominal rotation speed
WeightInlet flange
Exhaust flange

Ordering Information
60 Hz: 285 l/min (10 cfm)
60 Hz: 8.2 cfm

60 Hz: 0.55 kW
60 Hz: 1800 rpm

50 Hz: 237 l/min (14.2 m3/h)
50 Hz: 11.6 m3/h

10-4 mbar
2 x 10-3 mbar
2 x 10-2 mbar
20 mbar
160 g/h0.25/0.6 l

25 kg (55 lbs)
25KF DN
25KF DN

Dual Stage Rotary Vane Pump		
DS 302 with 1 phase worldwide motor*

Part Number
9499325

Oil and Accessories		
Rotary vane fluid, DS19 type, 1 liter		Rotary vane fluid, DS19 type, 1 liter (USA)*		
Rotary vane fluid, DS19 type, 1 gallon (USA)		
Oil mist eliminator		Oil mist replacement cartridge (pkg. of 2)		NW 25 oil exhaust filter		
NW 25 oil exhaust replacement cartridge		
Oil return kit		Oil drain extension		
European plug power cable 2 meters 1 ph		
USA plug power cable 2 meters 1 ph		
UK plug power cable 2 meters 1 ph		
NW 25 centering ring Viton		NW 25 aluminum clamp		

* 1 phase motors (100-120 / 200-230) V ±10%, 50/60 Hz.
All motors comply with CE and UL/CSA standards.

50 Hz: 0.45 kW
50 Hz: 1500 rpm

Minor Maintenance Kit		
Contains all the valves, O-rings and seals
to refurbish the pump to vacuum integrity

Part Number

Major Maintenance Kit		
Includes all the items of the minor kit plus the vanes

Part Number
94993819499370* According to PNEUROP 6602

Part Number
9499390K7516301
K7516302
9499395
9499394
9499392
9499342
9499376
9499375
9499396
9499397
9499398
KC25AV
KQ25AWP* When these pumps are used in Leak Detectors applications, werecommend the use of Rotary Vane Fluid (Elite-Z mechanical),P/N 695409005 as it features a lower vapor pressure.

14

15

ROTARY VANE PUMPSAGILENT ROTARY VANE PUMP MODELS► Agilent DS 402

DS 402 - Pumping Speed Curve(Torr)
10-3

10-2

10-1

1

101

50

Rotary Vane
Pumps

100

6 (0.24)40 (1.57) 40 (1.57)

242 (9.5)

242 (9.5)

6 (0.24)40 (1.57) 40 (1.57) 40 (1.57) 40 (1.57)

130 (5.1) 130 (5.1)

261 (10.3) 261 (10.3) 100 (3.9) 100 (3.9)

164 (6.5) 164 (6.5)

541 (21.3) 541 (21.3)

Effective Pumping Speed 50 Hz

230 (9.0) 230 (9.0)

10

10

5

(m3/h)

(cfm)
1
1

0.5

0.1
0.1

10 -3

10 -2

10 -1

1

10 1

10 2

(mbar)
Pressure

Technical Specifications
Free air displacement
Pumping speed*
Ultimate partial pressure*
Ultimate total pressure*
Ultimate total pressure with gas ballast*
Water vapor tolerance
Water vapor capacity
Oil capacity min/max
Motor rating 1ph
Motor rating 3ph
Nominal rotation speed
WeightInlet flange
Exhaust flange

Ordering Information
60 Hz: 410 l/min (14.5 cfm)
60 Hz: 12.3 cfm

60 Hz: 0.55 kW
60 Hz: 0.90 kW
60 Hz: 1800 rpm

50 Hz: 342 l/min (20.5 m3/h)
50 Hz: 17.4 m3/h

10-4 mbar
2 x 10-3 mbar
1 x 10-2 mbar
30 mbar
350 g/h0.5/1 l35 kg (77 lbs)
25KF DN
25KF DN

50 Hz: 0.75 kW
50 Hz: 0.75 kW
50 Hz: 1500 rpm

Dual Stage Rotary Vane Pump		
Part Number
DS 402 with 1 phase worldwide motor*
9499330
DS 402 with 3 phase worldwide motor**
9499331* 1 phase motors (100-120 / 200-230) V ±10%, 50/60 Hz.
** 3 phase motors (200-220 / 380-415) V ±10% at 50 Hz or(200-230 / 460) V ±10% at 60 Hz.
All motors comply with CE and UL/CSA standards.
Minor Maintenance Kit		
Contains all the valves, O-rings and seals
to refurbish the pump to vacuum integrity

Part Number

Major Maintenance Kit		
Includes all the items of the minor kit plus the vanes

Part Number
9499382

Oil and Accessories		
Rotary vane fluid, DS19 type, 1 liter		Rotary vane fluid, DS19 type, 1 liter (USA)*		
Rotary vane fluid, DS19 type, 1 gallon (USA)		
Oil mist eliminator		Oil mist replacement cartridge (pkg. of 2)		NW 25 oil exhaust filter		
NW 25 oil exhaust replacement cartridge		
Oil return kit		Oil drain extension		
European plug power cable 2 meters 1 ph		
USA plug power cable 2 meters 1 ph		
UK plug power cable 2 meters 1 ph		
NW 25 centering ring Viton		NW 25 aluminum clamp		

9499371* According to PNEUROP 6602

Part Number
9499390K7516301
K7516302
9499395
9499394
9499392
9499342
9499376
9499375
9499396
9499397
9499398
KC25AV
KQ25AWP* When these pumps are used in Leak Detectors applications, werecommend the use of Rotary Vane Fluid (Elite-Z mechanical),P/N 695409005 as it features a lower vapor pressure.

16

17

ROTARY VANE PUMPSAGILENT ROTARY VANE PUMP MODELS► Agilent DS 602

DS 602 - Pumping Speed Curve(Torr)
10-3

10-2

10-1

1

101

50

Rotary Vane
Pumps

100

6 (0.24)40 (1.57) 40 (1.57)

242 (9.5)

242 (9.5)

6 (0.24)40 (1.57) 40 (1.57) 40 (1.57) 40 (1.57)

130 (5.1) 130 (5.1)

261 (10.3) 261 (10.3) 100 (3.9) 100 (3.9)

164 (6.5) 164 (6.5)

541 (21.3) 541 (21.3)

Effective Pumping Speed 50 Hz

230 (9.0) 230 (9.0)

10

10

5

(m3/h)

(cfm)
1
1

0.5

0.1
0.1

10 -3

10 -2

10 -1

Alternative mounting holes are also available, with thread 1/4-20UNC

1

10 1

10 2

(mbar)
Pressure

Dimensions: millimeters (inches)Technical Specifications
Free air displacement
Pumping speed*
Ultimate partial pressure*
Ultimate total pressure*
Ultimate total pressure with gas ballast*
Water vapor tolerance
Water vapor capacity
Oil capacity min/max
Motor rating 1ph
Motor rating 3ph
Nominal rotation speed
WeightInlet flange
Exhaust flange

Ordering Information
60 Hz: 605 l/min (21.4 cfm)
60 Hz: 17.6 cfm

60 Hz: 0.90 kW
60 Hz: 0.90 kW
60 Hz: 1800 rpm

50 Hz: 504 l/min (30.2 m3/h)
50 Hz: 25 m3/h

10-4 mbar
2 x 10-3 mbar
1 x 10-2 mbar
30 mbar
550 g/h0.5/1 l35 kg (77 lbs)
25KF DN
25KF DN

50 Hz: 0.75 kW
50 Hz: 0.75 kW
50 Hz: 1500 rpm

Dual Stage Rotary Vane Pump		
Part Number
DS 602 with 1 phase worldwide motor*
9499335
DS 602 with 3 phase worldwide motor**
9499336* 1 phase motors (100-120 / 200-230) V ±10%, 50/60 Hz.
** 3 phase motors (200-220 / 380-415) V ±10% at 50 Hz or(200-230 / 460) V ±10% at 60 Hz.
All motors comply with CE and UL/CSA standards.
Minor Maintenance Kit		
Contains all the valves, O-rings and seals
to refurbish the pump to vacuum integrity

Part Number

Major Maintenance Kit		
Includes all the items of the minor kit plus the vanes

Part Number
9499382

Oil and Accessories		
Rotary vane fluid, DS19 type, 1 liter		Rotary vane fluid, DS19 type, 1 liter (USA)*		
Rotary vane fluid, DS19 type, 1 gallon (USA)		
Oil mist eliminator		Oil mist replacement cartridge (pkg. of 2)		NW 25 oil exhaust filter		
NW 25 oil exhaust replacement cartridge		
Oil return kit		Oil drain extension		
European plug power cable 2 meters 1 ph		
USA plug power cable 2 meters 1 ph		
UK plug power cable 2 meters 1 ph		
NW 25 centering ring Viton		NW 25 aluminum clamp		

9499371* According to PNEUROP 6602

Part Number
9499390K7516301
K7516302
9499395
9499394
9499392
9499342
9499376
9499375
9499396
9499397
9499398
KC25AV
KQ25AWP* When these pumps are used in Leak Detectors applications, werecommend the use of Rotary Vane Fluid (Elite-Z mechanical),P/N 695409005 as it features a lower vapor pressure.

18

19

AGILENT HS SERIES PUMPSFEATURES AND BENEFITS► Exhaust Filters - Oil Mist Eliminator ► Foreline Roughing Traps
The exhaust filters (below, left) and the oil mist eliminator(right), retain the oil vapors which would otherwise be
expelled into the atmosphere during pumpdown and gas
ballast operation.
Some models feature an oil return line to allow condensed oilor fluid to return to the pump reservoir.

Agilent’s new traps are designed to prevent the backstreamingof mechanical pump fluids. Copper and stainless steel gauze
inserts are designed to reduce oil backstreaming. Molecularsieve inserts are available for applications where it is desirable
to increase water vapor pumping speed while eliminatingbackstreaming above the trap.

ROTARY VANE PUMPS● Agilent’s HS 452 and HS 652 rotary vane
pumps employ an innovative frequency inverter
technology that delivers optimal and consistent
performance throughout the worldwide range ofvoltage and frequency conditions.
● Operating with low power requirements, themicroprocessor-controlled frequency inverter,
combined with a 3-phase motor, is an efficientdriving unit capable of delivering the high starting
torque required for a dual-stage oil pump.
● Green technology: environmentally friendly thanks
to reduced power requirements, low start up
current, minimum oil mist at pump exhaust.

Rotary Vane
Pumps

AGILENT ROTARY VANE PUMP ACCESSORIES

I/O and RS232/RS485 Communication• Adjustable pumping speed from 45 to 68
Hz permits easy integration and reduces
noise levels.

Ordering Information
NW25 oil exhaust filter (left)
Oil mist eliminator DS 102 - DS 602 (right)• Pump performance can be tailoredo specific applications to reduces
system costs.

Ordering Information
PN 9499392
PN 9499395

Foreline/Roughing Traps: please call Agilent for ordering information

Universal Input Voltage

For any further details on our rotary vane pump accessories,
please contact Agilent Vacuum Products.

• Truly universal single-phase voltage andfrequency provide worldwide compatibility.
• Constant performance at different input
frequencies.

Remote Diagnostics

Agilent Oil, GP Type Mechanical Pump Fluid
Agilent GP Type Fluid is a mechanical pump fluid
recommended for use in non-corrosive applications. As aresult of molecular distillation, it has low vapor pressure and,
therefore, backstreams less than undistilled refinery products.

• Remote monitoring and control of oil
consumption, power and current.
• Reduces maintenance costs, improves
uptime and offers higher reliability.

Ordering Information
Description
1 liter bottle
1 gallon bottle

Part Number
K7516301
K7516302

Reduced Power Requirements

Shipping Weight lbs (kg)3.5 (1.6)
14.0 (6.4)• Inverter technology reduces the power
required compared to traditional single
phase motors.
• Circuit-breakers are no longer required,resulting in reduced system costs.

Higher Nominal Rotational Speed(2000 rpm)
• The pump is ideally suited for steady andhigh gas-load applications.

20

21

ROTARY VANE PUMPSAGILENT ROTARY VANE PUMP MODELS
HS 452 - Pumping Speed Curve

KF25 EXHAUST FLANGE

40 (1.6)271 (10.7)

245 (9.6)(m3/h)

199 (7.8)

164 (6.5)

261 (10.3)

The first rotary vane pumps with truly “smart” capabilities.
• Agilent’s HS 452 and HS 652 pumps employ an innovative
frequency inverter technology that delivers optimal andconsistent performance while encompassing the worldwide
range of voltage and frequency conditions.
• Operating with low power requirements, themicroprocessor controlled frequency inverter, combinedwith a 3-phase motor, is an efficient driving unit capableof delivering the high starting torque required of a dualstage oil pump.
• HS 452 and HS 652 technology solves the common
problems inherent in traditional single-phase motors. SmartPumps start with inrush current about 7 times lower thanthat of equivalent traditional pumps.

Technical Specifications
Free air displacement
Pumping speed*
Ultimate total pressure*
Ultimate total pressure with gas ballast*
Operating voltage
Inverter maximum output power
Nominal rotation speed
Weight

101

102

10 3

50

5
1
-2

10

10-1

99 (3.9)• Unlike traditional pumps, Smart Pump’s software driven
startup procedure recognizes faulty pumps within seconds,
and so avoids uncontrolled load conditions. The result isvery easy pump integration.
• The 3-phase inverter output is constant and independent of
single-phase input frequencies and voltages. Motorefficiency and power factors remain optimal and as a result
motor and pump wo asy to drive the pump. Pump
performance can be tailored by setting the rotational speed
for specific applications. The software monitors and logs
pump parameters making it possible to perform pump andsystem diagnostics.

1

10515 (20.3)

Dimensions: millimeters (inches)(Torr)

101
10-3
130 (5.1)

10-1

Pressure (mbar)

KF25 INLET FLANGE

10-2

(cfm)

75 (2.9)

Pumping Speed

100

10-3

HS 452 - Pump Down Curve1(mbar)
Pressure

101

102

10 1
1
10 -1
10 -2

103

0

* 200 liters chamber

60

120

180

240

300

Time (sec)

Ordering Information
Pump		HS 452 Smart Pump 1 phase worldwide motor
IP44 accessory connector Kit		

Part Number
9499360
9499367

Accessories		
NW25 oil exhaust filter
Oil return kit
Oil drain extension
Rotary vane fluid, DS19 type, 1 liter
European plug power cable 2 meters 1 ph
USA plug power cable 2 meters 1 ph
UK plug power cable 2 meters 1 phMinor maintenance kit
Major maintenance kit

Part Number
9499392
9499376
9499375
94993909499396
9499397
9499398
9499371
9499382

Application Note
Steady state high gas load applications exploit the major
benefits of the smart technologies. When operated atpressures lower than the auto-tuning pressure, the HS 452 andHS 652 outperform the equivalent traditional DS 402 and DS
602 pumps by 10% at 60 Hz and by 30% at 50 Hz.
The Smart Pumps can run below auto-tuning pressure at theirfull speed of 2000 rpm. As shown in the pumping speed curve

at top right, the HS 452 has an auto-tuning pressure of 100
mbar while the HS 652 has an auto-tuning pressure of 40 mbar.
Despite the lower power requirements of 780 W max, theSmart Pumps deliver good performance in the roughing phase.
Fast cycling or inrush applications need to be evaluated on acase by case basis with our application engineers.

27 m3/h (16 cfm)
22 m3/h (13 cfm)
2 x 10-3 mbar
1 x 10-2 mbar
100-120/200-240 V ±10%, 50/60 Hz
780 W
2000 rpm
33 kg (73 lbs)

* According to PNEUROP 6602

22

10 2

Rotary Vane
Pumps

► Agilent HS 452

23

ROTARY VANE PUMPSAGILENT ROTARY VANE PUMP MODELS
HS 652 - Pumping Speed Curve

KF25 EXHAUST FLANGE

40 (1.6)271 (10.7)

245 (9.6)(m3/h)

199 (7.8)

164 (6.5)

261 (10.3)

The first rotary vane pumps with truly “smart” capabilities.
• Agilent’s HS 452 and HS 652 pumps employ an innovative
frequency inverter technology that delivers optimal andconsistent performance while encompassing the worldwide
range of voltage and frequency conditions.
• Operating with low power requirements, themicroprocessor controlled frequency inverter, combinedwith a 3-phase motor, is an efficient driving unit capableof delivering the high starting torque required of a dualstage oil pump.
• HS 452 and HS 652 technology solves the common
problems inherent in traditional single-phase motors. SmartPumps start with inrush current about 7 times lower thanthat of equivalent traditional pumps.

Technical Specifications
Free air displacement
Pumping speed*
Ultimate total pressure*
Ultimate total pressure with gas ballast*
Operating voltage
Inverter maximum output power
Nominal rotation speed
Weight

101

102

10 3

50

5
1
-2

10

10-1

99 (3.9)• Unlike traditional pumps, Smart Pump’s software driven
startup procedure recognizes faulty pumps within seconds,
and so avoids uncontrolled load conditions. The result isvery easy pump integration.
• The 3-phase inverter output is constant and independent of
single-phase input frequencies and voltages. Motorefficiency and power factors remain optimal and as a result
motor and pump working temperatures remain low andconstant worldwide.
• Smart Pumps can be remotely driven via discrete I/O or
RS232/RS485 interfaces. Agilent’s T-plus Navigator
Software facilitates communication making it very easy todrive the pump. Pump performance can be tailored by
setting the rotational speed for specific applications.
The software monitors and logs pump parameters making
it possible to perform pump and system diagnostics.

1

10515 (20.3)

Dimensions: millimeters (inches)(Torr)

101
10-3
130 (5.1)

10-1

Pressure (mbar)

KF25 INLET FLANGE

10-2

(cfm)

75 (2.9)

Pumping Speed

100

10-3

HS 652 - Pump Down Curve1(mbar)
Pressure

101

102

10 1
1
10 -1
10 -2

103

0

* 200 liters chamber

60

120

180

240

300

Time (sec)

Ordering Information
Pump		HS 652 Smart Pump 1 phase worldwide motor
IP44 accessory connector Kit		

Part Number
9499365
9499367

Accessories		
NW25 oil exhaust filter
Oil return kit
Oil drain extension
Rotary vane fluid, DS19 type, 1 liter
European plug power cable 2 meters 1 ph
USA plug power cable 2 meters 1 ph
UK plug power cable 2 meters 1 phMinor maintenance kit
Major maintenance kit

Part Number
9499392
9499376
9499375
94993909499396
9499397
9499398
9499371
9499382

Application Note
Steady state high gas load applications exploit the major
benefits of the smart technologies. When operated atpressures lower than the auto-tuning pressure, the HS 452
and HS 652 outperform the equivalent traditional DS402 andDS602 pumps by 10% at 60 Hz and by 30% at 50 Hz.
The Smart Pumps can run below auto-tuning pressure at theirfull speed of 2000 rpm. As shown in the pumping speed curve

at top right, the HS 452 has an auto-tuning pressure of 100
mbar while the HS 652 has an auto-tuning pressure of 40 mbar.
Despite the lower power requirements of 780 W max, theSmart Pumps deliver good performance in the roughing phase.
Fast cycling or inrush applications need to be evaluated on acase by case basis with our application engineers.

40.3 m3/h (23.8 cfm)
32 m3/h (19 cfm)
2 x 10-3 mbar
1 x 10-2 mbar
100-120/200-240 V ±10%, 50/60 Hz
780 W
2000 rpm
33 kg (73 lbs)

* According to PNEUROP 6602

24

10 2

Rotary Vane
Pumps

► Agilent HS 652

25

ROTARY VANE PUMPSMS 40+ FEATURES AND BENEFITS● Sophisticated Electronics and excellent mechanical design allow highpumping capacity and reduced dimensions.
● Suitable for steady and high gas throughput conditions at pressures
below 10 Torr, in applications like Mass Spectrometry (including LC-MS,
ICP-MS, GC-TOF-MS, etc.), Electron Microscopy, and other Scientific
Instrumentation.

Inverter Technology

Highest Performances / Size Ratio• Proven on-board electronics allows
constant performance worldwide• Base pressure below 5 x 10-2 mbar
(3.75 x 10-2 Torr)• Truly universal voltage and frequency• High pumping speed over a wide range,from atmosphere to 1 mbar (0.75 Torr)• Single phase

Please contact Agilent to qualify use in cyclic applications.

• Inverter technology allows to manage
pumping speed to more than 45 m3/h
(26.5 cfm)

• Inverter electronics enable tailoring
pumping speed to each application• Best noise level, with the highest
throughput

Interface Capabilities• I/O and RS232/RS485 enable adjustmentof operating parameters simplifying
system integration• Remote diagnostic allow thecontrol of:- Oil level- Temperature- Power- Current

297.25 [11.70]
127.00 [5.00]

89.00 [3.50]

246.50 [9.70]

T-Plus Software

Worldwide Service Capability

Smallest Dimensions

Green Technology• T-Plus software allows control of pumpparameters via PC, improving uptime
and reliability, and reducing
maintenance costs• Three levels of Product Support- 24h Advance Exchange- Factory Repair- Upgrade Program
allow global coverage of service needs tomaximize productivity and uptime• The smallest single stage 40 m3/h pump in
the market. 297 x 418 x 225 mm(11.69 x 16.46 x 8.86 inches)• Environmentally friendly andcost-effective due to reduced power
requirements, low start up current(< 10 A) and stand by mode

418.48 [16.48]

224.97 [8.86]

13.50 [0.53]

199.00 [7.83]

117.00 [4.61]• Including on-board electronics, anti suckback valve, integrated exhaust filter andoil return kit
73.50 [2.89]

64.50 [2.54]

113.00 [4.45]

130.50 [5.14]• Easier system integration• Simple maintenance• Allows smaller overall instrument design,
therefore reducing costs
26

27

Rotary Vane
Pumps

Mono Stage Rotary Vane Pump
Small Footprint, High Pumping Capacity

ROTARY VANE PUMPSAGILENT ROTARY VANE PUMP MODELSMS 40+ Pumping Speed Curve

297.25 [11.70]
127.00 [5.00]

89.00 [3.50](Torr)
-2

-1

10

100

10

101

1

102

50

10

10

418.48 [16.48]
117.00 [4.61]

13.50 [0.53]

Speed

246.50 [9.70]

Rotary Vane
Pumps

► Agilent MS 40+

5(m3/h)

(cfm)
1

0.1
0.1

73.50 [2.89]
64.50 [2.54]

113.00 [4.45]

0.5

224.97 [8.86]

199.00 [7.83]

1

10 -2

10 -1

10 1

1

10 2

10 3

(mbar)

130.50 [5.14]

Pressure
Dimensions: millimeters (inches)1200 rpm

1500 rpm

1800 rpm

Technical Specifications
		9499225
Free air displacement
828 l/min (29.2 cfm; 49.7 m3/h)
Pumping speed (at 5 mbar inlet pressure)
40 m3/hUltimate total pressure*
5x10-2 mbar
Oil capacity min/max
1l
Motor rating 1ph 		0.75 kW
Noise level with gas ballast closed≤ 62 dB(A)Oil temperature (pump operating)60 °C
		140 °F
IP Value			
Installation category		
Pollution degree			
Operating temperature range
12-40 °C
Nominal rotation speed
1450 rpm
Weight		33 kg (73 lbs)
Inlet flange		
25KF DN
Exhaust flange		
25KF DN
Dimensions: - length
418 mm- width297 mm- height
228 mmNominal Input Voltage
200-240 V
Input frequency		
50/60 HzMax input power		1200 VAInternal Main Fuse (TT type)12.5 A

9499240828 l/min (29.2 cfm; 49.7 m3/h)
40 m3/h5x10-2 mbar
1l0.75 kW≤ 62 dB(A)60 °C
140 °F20II2
12-40 °C
1450 rpm
33 kg (73 lbs)
40KF DN
25KF DN
418 mm
297 mm
228 mm200-240 V
50/60 Hz
1200 VA
12.5 A

9499241828 l/min (29.2 cfm; 49.7 m3/h)
40 m3/h5x10-2 mbar
1l0.75 kW≤ 62 dB(A)60 °C
140 °F

Ordering Information
Dual Stage Rotary Vane Pump		
MS 40+ RVP, 25KF Inlet Flange,
with I/O and RS232/485 Interface,
without Oil Level Sensor		
MS 40+ RVP, 40KF Inlet Flange, Full Optionalwith I/O and RS232/485 Interface,with Oil Level Sensor		
MS 40+ RVP, 40KF Inlet Flange, Base Version		

Part Number

Oil and Accessories		MS 40+ Exhaust filter		
MS 40+ 1 Litre Oil Tank		
MS 40+ Maintenance Kit		
Power cable EU		
208Vac US Power cable		Power cable UK		Power cable IEC320		
T-PLUS Navigator SW (w/serial cable)		

9499225
9499240
9499241

12-40 °C
1450 rpm
33 kg (73 lbs)
40KF DN
25KF DN
418 mm
297 mm
228 mm200-240 V
50/60 Hz
1200 VA
12.5 A* According to PNEUROP 6602 - No gas ballast port

28

29Part Number
9499201
9499202
9499203
94993969499400
9499398
9499399
9699883