LE-3500R
Multi Protocol Analyzer
LE-2500R
Battery-powered portable communication
analyzer with Wi-Fi* for remote measurement.

MULTI PROTOCOL ANALYZER

LE-3500R
● 210
（W）
×154
（D）
×38
（H）
mm,
Approx. 760 g

ModbusSupported

MULTI PROTOCOL ANALYZER

LE-2500R
● 210
（W）
×154
（D）
×38
（H）
mm,
Approx. 760 g

Standard
communications ＞＞＞
Expanded
communications ＞＞＞

TTL/UART/I2C/SPI

RS-232C

RS-422/485

CAN/LIN

CC-Link Current Loop

RS-530/X.20/X.21/V.35＊Wi-Fi function is available only in Japan, USA, and Canada.

Battery-powered portable communication analyzer.
MULTI PROTOCOL ANALYZER

LE-3500R / LE-2500R

LINEEYE released the first LE-series model in 1986. Since then, LINEEYE has been developing the LE Series in
response to the requirements of a large number of customers, and now come to the fruition of the LE-3500R andLE-2500R which support remote measurement by Wi-Fi in the IoT era. The LE-3500R and LE-2500R incorporate
versatile analysis functions and excellent portability, thus fully supporting the trouble analysis of communications
systems, industrial equipment, and a variety of in-vehicle networks as well as development tests and after-sale services.
TTL measurement port / External signal I/O terminal
RS-422/485 (RS-530) measurement port
RS-232C (V.24) measurement port
DC input connector

Exchangeable measurement board

AUX (RS-232C) connector
Elastomer resin

Power supply
switchUSB2.0 B connector(to connect with a PC)

Contrast
adjustment
volumeUSB2.0 A connector(to connect with a
USB flash)SD card slot

Special data items are expressed in individual signs.

Menu keyLine state LED(Example of display signs)
HDLC/SDLC flag
Short frame

Measurement
start key

Idle time (frame interval time)

Block check : Good
Block check : Error
Parity error

Time stamp (frame arrival time)

Framing error

Continuously displays the state ofcommunications line with a 2-color LED.

In RS-232C
Red: ON space state
communications
Green: OFF mark state
OFF: Not in use or not wired

PE and FE simultaneous error
BreakSPEED In half-duplex mode
Max speed
4Mbps

4M bps100MB
MemoryMax speed
2Mbps

2M bps64MB
MemoryMax speed
1Mbps

1 M bps
Max speed
500Kbps

500Kbps

16GB
USB Flash Drive/
SD card support

8GB
CF card support

MULTI PROTOCOL ANALYZER
COMPACT PROTOCOL ANALYZER

LE-1500
An entry model dedicated to Async/PPP which has asufficient measurement function with moderate price.

32GB
USB Flash Drive/
SD card support

64MB
Memory2.4MB
MemoryLE-2500R

128GB
CF card support

MULTI PROTOCOL ANALYZER

LE-3500R
A perfect model incorporating statisticalanalysis and program simulation functions.

USB Flash Drive(LE-8200A)

MULTI PROTOCOL ANALYZER

LE-8200A
LE-8200
The high-end model boasting plentiful performancewith a large-sized color display and versatilefunctions.
*See the LE-8200(A) catalog for details.

A high cost-performance multi-protocol
model with expandability.

*See the LE-1500 catalog for details.

Async, PPP

Multi Protocol（Async, Sync, BSC, SDLC,HDLC, X25, PPP, I2C, CAN, LIN ････）

Measures UART/I2C/SPI without using optional board.
It has a measurement port for TTL(1.8V – 5V) as its standard port. Measurement tests for sensor,
LAN and wireless modules of I2C, SPI, and UART interface, for AD conversion IC, or for memory
IC are available without changing the measurement board.
[ A 5-wires TTL prove cable is attached ]TTL line：Terminals for UART, I2C, and SPI
TRG line：External trigger input/output, PWR terminal
Note: The voltage of selected signal level(1.8 / 2.5 / 3.3 / 5V) is output from the PWR
terminal.
Pin header connector of 2.54mm pitch, equivalent
to HIF3FC-10PA-2.54DS(71)(HIROSE ELECTRIC
CO., LTD.)It can be used for TTL communication
signal and external trigger signal. If
you want to use both of them at thesame time, purchase one more thisprove cable.

[ I2C monitor display example ][ Interface setting ][ SPI monitor display example ]Supports RS-232C of DSUB 9 pin

Detachable terminal for RS-422/485

A monitor cable for DSUB25 pin, a monitor cable for DSUB9 pin, and a changer
for DSUB25 pin to DSUB9 pin are attached. You can connect devices of
DSUB25 pin or DSUB9 pin without any optional cables.

RS-422/485 cable can be connectedwith the analyzer directly. As it isdetachable, you do not need to unscrew
the cables, when disconnecting theanalyzer from the monitoring line.

Mega Speed Measurement

Remote Control by Wi-Fi *1

Analysis is possible at any baud rate [Baud rate setting display]from low speed to high speed.(*)
Margin tests on communication
speed deviation are simple.

You can analyze the log data or convert the data into text file on your PC by connecting
the PC in which the PC link software is installed with protocol analyzer(s) by USB or
Wi-Fi. You can also capture the screen image of analyzer in the Android smart phone
display to send it by email by connecting the smart phone with the analyzer by Wi-Fi.

* Using high precision DPLL technology
for open baud rate support, transmission
and reception speeds can be separately
set to an effective 4 digits.

*1 The light edition of the PC link software for Windows PC “LE-PC300R” is attached.The application forAndroid smart phone “LE-REMOTE4” can be downloaded from Google Play.Wi-Fi function is available
only in USA, Canada, and Japan.
STA mode

Auto Save/Long Recording Time
AccessPoint

You can record communications data into the 64MB memory endlessly or stop
recording it automatically when the memory is full. Furthermore, an auto savefunction makes it possible to save the monitored content of captured memory into a
SD card or a USB flash. Auto Save continuously saves data into the measurement
log of a user-specified file size, using ring recording as long as the card has space.
It is useful for identifying rare communication failures of unknown cause
Target line Continuous recording time reference(LE-3500R)
speed (bps)Main memory only
When using 32GB SDHC card
9600

Approx. 3.5 hrs

Approx. 80 days

115.2K

Approx. 18 min

Approx. 6.5 days

1Ｍ

Approx. 140 sec

Approx. 20 hrs*Calculated for full-duplex transmission of 1,000 byte data frames per second.
If the interval of frames are longer, the recording time will be longer.

AP mode>>> See page 7■ Expands measurement targets
Measurement targets can be expanded by optional measurement boards or connection cables.

ＣＡＮ, LIN[ SD card / USB flash are supported ]Logic Analyzer
Measurement

andSignalVoltageCommunication line timing is analyzed and displayed as a logic analyzer
display to a time resolution of max. 50 ns. You can measure the voltage levelof RS-232C or TTL signals by the signal voltage measurement function.
[Example of logic analyzer measurement]

When the optional board OP-SB7GX is inserted.

RS-530[Example of signal voltage measurement]

When the optional board OP-SB10N is inserted.

>>> See page 8 - 9

A monitor function to visualize communications data.
Supports multi-protocols
The online monitor featurerecords communications data inthe capture memory and provides
an easy-to-understand display
for the type of protocol, without
affecting the communications
line. As a standard feature,
LINEEYE protocol analyzers
support various communications
standards from asynchronoustopacketcommunication.
Depending on the test, you canselect bit transfer sequence andpolarity, as well as modulation
format from NRZ, NRZI, FMO,
FM1. The feature allows to
support effective analysis byomitting SYN codes and using
SDLC/HDLC address filter.

Trigger Feature for Catching User-specified Events
The trigger feature allows you
to specify a communications
event as the trigger conditionand has measurement operations
executed automatically whenthat condition is satisfied. Up tofour pairs of conditions andoperations can be set, which is
helpful towards identifying
frequent intermittent faults that
occurs with communications
systems. And, the operation ofa trigger condition can be
specified as the condition foranother trigger, making it
possible to analyze complicatedoperations based on sequential
triggers.

[Data display][Example X.25 protocol translation][Example PPP translation][Trigger setup summary display][Example trigger condition setup][Example trigger action setup]

Line state LED[Example of connection for online monitoring]monitor cableCommunications
network
DCE

DTE

SD
RD

Beforeconnection

SD RD

After
monitorcable
coccection

Communications line state is indicatedin real time using 2-color LEDs

LINEEYE protocol analyzers
can analyze communications
data and automatically set
basic measuring conditions,such as communications speed,
character framing, data code,
synchronization
character,
BCC/FCS, etc. This is effective
for monitoring lines of unknown
communications conditions.

[Monitor condition auto setting – Search display][Monitor condition auto setting – Determination display]*The auto setting is not accurate withsmall volumes of communications
data or data that contains manyerrors.

Records Time Data with Communication Data
LINEEYE protocol analyzers
record not only communications
data but the time (time stamp)of transmissions and receptions
as well as idle time; therefore
failure time and timeout status
can be checked. It is also
possible to record the informationof changes in control lines atthe same time. For ASYNC/
BURST communications, setting
the idle time to be the frame end
is available in the range of 1 to
100ms.

Monitor Condition Auto Setting

Signal voltage measurement feature for RS-232C and TTL[Example record data selection]

You can measure the voltage(current
value,
maximum,
minimum) of 4 signal lines of
RS-232C or TTL measurement
port. You can also measure delay
time of control line change (suchas the change from RTS to CTS).

[Time stamp / idle time display][Example delay time measurement ]Only for LE-3500R

Statistical Analysis Capabilities
Statistics can be compiled fortransmission and reception data
sets, frames and the number of
established trigger events, andsubsequently displayed as a graph(Unit: 1-240 min.). This helps tounderstand communications
traffic and error frequency fora specific time period.

[Timing data display][Graph of statistically analyzed data]

BERT function to measure the occurrence rate of communications errors.
BERT function enables you tomeasure transmission quality ofcommunications lines by aloop-back or interactive connection.
It is possible to measure evaluation
parameters (bit error count,block error count) conforming to
ITU-T G.821 Notification, hence
enabling bit error rate evaluations
and fault point identification.
Elaborate test patterns andfunctions such as bit error forced
interrupt are comparable todedicated equipment.

Evaluation is possible in ASYNC
or SYNC mode, by specifying
measurement period (continuous,received bits, specified time,
repeat) or test pattern.

[Example connection for BERT]
Loop-back test

Test
patternCommunications
network

Interactive test

Test
patternCommunications
network

Testpattern

Loopback
point

Once started, the results ofmeasured line quality are displayed
and updated in real time. Whenfinding the error bit, LINEEYE
analyzers can output the external
trigger to inform other equipments.
Repeat mode allows you to know
the error rate for the specific time
range in the communication lines.

[BERT setting display][Example BERT measurement][Contents of BERT measurement]
Savail Available measurement in seconds
R-Bit

Effective bits received

E-Bit

Error bit count0〜9999999

Loss

SYNC loss count0〜9999999〜9.99E9

R-Blk

Effective blocks received 0〜9999999〜9.99E9
Block error count0〜99990〜9999999〜9.99E9

E-Blk

Bit-ER Bit error rate

0〜9.99E-9〜1

Blk-ER Block error rate

E-Sec Error in seconds

0〜9999

%E.F.S Normal operation rate 0.000〜100.000%

0〜9999999〜9.99E9
0〜9.99E-9〜1

MULTI PROTOCOL ANALYZER

LE-3500R / LE-2500R

Online Monitoring,
Simulation and
BERT Support All-in-One
Must-haveMust-have measurement
measurement tool
tool forfor on-site
on-site test
test oror communication
communication trouble
trouble analysis.
analysis.
Simulation function to conduct transmission and reception tests in place of target equipment under test.
With the simulation feature, the LINEEYE protocol analyzers act as the counterpart tothe target device and perform transmission and reception tests according to protocol.
Even in the early stages of development when matching devices are not available,
tests can be run at near to actual operating status. After checking the communications
protocol step by step in LINEEYE[Control lines and data transmission timing]
analyzer’s own original MANUAL
mode, a developer can create a
simple program to branch
conditions using menu selection
and test more complicatedcommunications
protocols.
Communications speed can be[Example of connection for simulation]
freely set ; therefore margins can beevaluated by intentionally shifting
communications speed, and error
Test data
response processing can bechecked using test data thatmixes in data with parity errors.
In addition, data transmission
can be linked with the changes
in the signal lines such as RTS
and CTS at the preset timing.

■ POLLING mode
The POLLING mode simulates the slave and master units in multidrop (1:Nconnection) polling protocols. In the slave mode, the LINEEYE protocol analyzers
check the number of received[Polling setup display]frames that are assigned theiraddress and whether errors
occur or not, replying withuser-set data. In the mastermode, they send polling
messages to 32 slave units, andcheck and display replies fromeach slave.
Only for LE-3500R■ BUFFER mode

In the BUFFER mode, you can select between transmission and reception, and sendtransmitted or received data that has been captured in the buffer using the unit's
monitoring feature, as simulation data without requiring further manipulation. Thismode is effective in conducting[Buffer data reproduction setup display]
reproducibility tests using thesame data as that monitored
under actual communications
conditions.

■ MANUAL mode
The MANUAL mode allows you to send the data registered in transmission table whichcorresponds to the "0" to "F" keys (10 groups, 16 types for each). The data can be sentwith one press of a key. While checking replies from a unit under development with themonitoring feature, you can easily and simply test the communications process. You
can also send fixed data by[Transmission table setup display]registering it under a key
combination of the SHIFT and"0" to "D" keys, as well as turn
RTS/CTS and DTR/DCD signallines on/off with the SHIFT and"E"/"F" key combinations.

■ FLOW mode
Flow control can be simulated on the transmission and reception-lines using X-on/off
flow control or the control line handshake. In the transmission mode, up to 16 cyclesof data from transmission start until a generated interrupt request can be displayed. In
the reception mode, you can set[Flow control test setup display]
the number of received data
cycles until a transmissioninterrupt request is generated,
as well as the time until thetransmission resume request is
generated.

■ ECHO mode
In the ECHO mode, LINEEYE protocol analyzers internally return received data. Buffer
echo to send back data by a reception frame, character echo to send back data by acharacter and loop back echo
[ECHO mode setup display]that simply loops back data canbe selected. It is used to test
displayterminals
andcommunications terminals.

Only for LE-3500R■ PROGRAM mode
By creating a purpose-specificcommandprogram,thecommunications protocol can be
flexibly simulated alongsidecondition monitoring. The programis created using the menu
selection, so it is easy to master.

[Program example][PROGRAM mode commands]
CommandOperation

SEND CHR□□□□□□□□

Sends max. 8 data sets.

SEND REG□

Sends data registered in transmission table under
specified REG No.

SEND BRK
WAIT

CHR

WAIT

FRM

WAIT

TM

Sends break signals (ASYNC only).

□□□□□□□□

Waits until receiving specified data (max. 8 data sets).
Waits until receiving 1 frame.

□□□□

GOTO L□□□

Waits for specified amount of time.
Jumps to specified label No.

CALL

L□□□

IF

CHR

□□□□□□□□IF

LN□＝□ L□□□

SET

REG

□

□□□□□□

Sets or increases/decreases value of specified REG
No.

SET

TM

□

□□□□□□

Controls specified timer and sets to specified value.

INT

TRG 0 L□□□

Jumps to subroutine of specified label No.

L□□□ Branches if specified data in reception buffer.
Branches if interface line is specified logic.

Interrupts specified label when trigger 0 condition issatisfied.

Easy-to-Use Handy Functions
Continue to Evolve
Firmware That Evolves

PC-compatible File Management Specification

The latest firmware with additional functions and improvements can be foundon our website. If you download it with your PC, you can then update to thelatest version via a serial/USB cable.

Test conditions and results such as measured data can be saved on optional SD
cards or USB flash in the files management format compatible with your PC. Of
course, files can be interchangeably used(*) between models. Therefore,
measurement data can be saved on-site with the LE-2500R, and analyzed ormanipulated in greater detail using the LE-3500R back in the office.

Menu-based Simple Operation* The LE-8200(A)/3500R/2500R/3500/2500/1500 are compatible in measurement data file. Part of files ordata saved in higher hierarchy models or new models, however, may not be available to lower hierarchymodels or conventional models.

Anyone can easily use LINEEYE protocol analyzers owing to the easy menu
selection system handed down from earlier models.
[Top menu display (LE-3500R)]

Types, names, sizes and thedate/time of files saved in thememory card can be checked.

[Example file operation display]

When many files have beensaved, the file filter featureallows you to specify the typeof file to be displayed.

[File filter setup display][Top menu display (LE-2500R)]

A backlit LCD makes it easy to view
measurement data at night and indark places.

Offline Analysis and Data Searches
Measurement data displays can be freely scrolled and paged. A powerfulsearch feature allows you to locate specific data and perform counting

Search key

Communications error (individual error type can be specified),communications data string of max. 8 characters (don’tcare and bit mask can also be specified), idle time beyond
a specified duration, specific time stamp (don’t care canalso be specified), external trigger matching data

Searchoperations

Find and display, counting

Auto RUN/STOP for Unmanned Measurement
By setting time and a date of measurement start and end, measurement can bedone automatically during the specified time period. For example,
measurement only for 3 hours from 18:00 to 21:00 every day is possible.
Besides, if the power ON auto run function is used, unmanned measurement
can be started automatically without pressing the RUN key after turning power ON.
[Example auto run display][Example search key setting]

Using “don’t care (*)”, you can search fortime stamp data from 10:30:00 to 10:39:59
as in this example.

Various Print Formats
Measured log data can be printed out in text format by the printer with indicated printing
range or can be saved into a SD card or USB flash by the auto save function. The log
file saved in a SD card or USB flash can be collectively* converted into a text file or a
CSV file by connecting the media with a PC and capture data into the PC using the PC
link software.
*The attached software (light edition, not the product version) can convert up to 3 files at the same time.

[Example statistical analysis printout][Example logic analyzer waveform printout][Printout setting display]

You can select the output
destination of the print
data (text data).

[Remote control setting display]

According to the environment, 【Utility image】
you can select the modes forConnection with the printer
Wi-Fi connection.

Continuous printing of logic analyzer
wave form and the result of statistics
analysis is available.

by AUX(RS-232C)

Dedicated printerDPU-414-PA

Wired connection
Connection with a PC
by USB or AUX(RS-232C)[Wi-Fi AP mode setting display]

PC with the PC link software

You can set the password
and radio channel for the
AP mode in which theanalyzer turns to be anaccess point.

Wi-Fi[Wi-Fi STATION mode setting display]

You can set the SSID andthe password to connect
the analyzer via anaccess point such as
Wi-Fi router.

You can control the analyzer remotelyfrom an Android smart phone andcapture the monitoring display to email
it by the smart phone.

You can confirm and edit the data by text
editor after converting it to text format.

LE-PC300R Enhances the Link
between Analyzers and your PC

PC Link Software

LE-PC300R●Cannot used with OP-SB7GX●The light edition of LE-PC300R is attached toLE-3500R/LE-2500R

Enables simultaneous control of multiple analyzers from a PC
The software supports the connection by USB, serial connection, and Ethernet
connection by optional converter of Ethernet and serial. For LE-3500R/2500R, it also
supports connection by Wi-Fi. It supports connection with multiple analyzers,
displays and converts the measured data in the external memory such as SD card.

【Linking multiple analyzers by USB, Wi-Fi and LAN】
Target line

Target line
PC

【USB/Serial connection, Wi-Fi connection, and External memory】
Target line

USB
LE-3500R

LE-2500R

Wi-Fi connection*1

SI-60FLAN（１０/１００BASE-TX）

Measurement
data

2

*
Wi-Fi router

Target line

Wired connection*1

Target line

PC with the PC link software
USB flashSD

LE-3500R
LE-2500R

LE-2500R

LE-3500R

SD cards*2*1: Connected by a USB cable or AUX cable (LE2-8V)*2: An interface to read the SD card is needed for PC.

*1: SI-60F is a converter of Ethernet and serial which is supported by LE-PC300R.
*2: A Wi-Fi router of IEEE802.11 b/g/n is available.

Allows the measurement data to be checked on your large PC screen.

▲ Normal display▲ HDLC translation display▲ Logic analyzer display

Records communication logs continuously on PC up to a maximum of 16GB

Converts the recorded data to text format or CSV format all at once

The remote monitor function allows to record the data measured by an analyzer on
the HDD or SSD.
The fixed buffer mode and ring buffer mode are available. The former stops
recording when the specified data size is reached, and the latter records data
endlessly within the limit of the specified size.
*Measurement time is limited to 10 minutes for the light edition of LE-PC300R.

Multiple files of communications logs can be converted to text or CSV format foruse on word processor or spreadsheet. Conversion to text is based on the print
format of the analyzer. In consideration of analysis on general search tool, it ispossible to delete decorative guides or time data, and to specify conversion of
sender or receiver data only.
* The number of the files which can be converted at the same time is limited to 3
for the light edition of LE-PC300R.

[Standard time for continuous recording on hard disk *1 ]Target line speed *2

When 1 GB is specified :(e.g.: 1 MB x 1,000 files)

When 16 GB is specified :(e.g.: 8 MB x 2,000 files)

9600 bps

Approx. 60 hrs

Approx. 960 hrs

19200 bps

Approx. 30 hrs

Approx. 480 hrs

38400 bps

Approx. 15 hrs

Approx. 240 hrs

Changes the System Language Automatically
The system language alternates automatically between English and Japanese
according to that of OS. This facilitates introduction of the software to development
bases outside Japan.

*1: In case of full-duplex communications line for transmission at 1 ms interval per 1 KB.
*2: Maximum communications speed that ensures recording of measurement data without failure will be about 1/5 of serialtransfer speed between analyzer and PC.

■LE-PC300R Specifications
Applicable analyzers

LE-3500R, LE-2500R, LE-3500, LE-2500, LE-1500

Analyzer connection

USB, Wi-Fi, Serial, and Ethernet(optional unit SI-60/SI-60F is needed.)

No. of analyzers to be connected

Multiple analyzers can be connected and controlled simultaneously.(No. of connectable analyzers depends on the performance of PC.)

Key emulation function

Presents the analyzer’s display on the PC screen to enable control in a manner as if operating the analyzer.

Measurement condition setting

Measurement conditions (communications parameters, trigger and simulation data) can be input and edited on the dedicated window on PC screen.

Remote monitor function

Recording modes Fixed buffer mode (Records data up to the specified size) or ring buffer mode (Records data endlessly whileleaving the latest data of the specified size) can be selected.

PC Link Software for CAN/LIN
For OP-SB7GX LE-PC7GX

Starts/stops measurement with analyzer, displays the measurement data on PC screen, and records data continuously.

Recording capacity Max. 16 GB can be specified up to 2,000 files in the unit of 1/2/4/8 MB data file.
Selectable among raw data, protocol translation and logic analyzer waveform.

Raw data
Display modes

Displays communications data accompanied by idle time, time stamp and line status.
Character code (10 kinds) and character size (small/medium/large) can be changed.

Protocol translation Translates and displays SDLC, X.25 and LAPD protocols. (Target protocols planned to be increased.)
Logic analyzer waveform Enlarges and reduces waveform, measures time between cursors, and rearranges signals.
Finds and displays the data that matches the search key.

Search function

Search key

Specified data string of max. 8 characters (don’t care and bit mask can also be specified), idle time
beyond a specified duration, specific time stamp (don’t care can also be specified), error (parity, framing,
BCC, break/abort, short frame: individual error type can be specified) external trigger matching data

Text-CSV conversion function

Specified number of recorded files can be converted to text or CSV format all together.

Bitmap conversion function
System requirements

Analyzer’s display shown by key emulation can be saved to bitmap files.

Composition

CD (Software) x 1, instruction manual x 1, user registration card x 1

Windows ® 7/8/8.1/10

This software links your PC and LE-3500R/2500R equippedwith CAN/LIN communications expansion kit OP-SB7GX.
It enables to analyze collected CAN/LIN data on your PC.
• USB, Serial, Ethernet, and Wi-Fi* to the PC• Key emulation function for remote control• Recode CAN/LIN data into the PC at maximum 16GB
• Display the specific ID frame at real time
• Data and timestamp search, text/CSV conversion• Set the analyzer conditions from the software• Read the measured file in the SD card or USB flash• OS: OS: Windows® 7/8/8.1/10*Version 2.00 or more of LE-PC7GX, OP-SB7GX, LE-3500R/2500R,
and the Wi-Fi connection environment are needed.

Expanded communications

TTL/I2C/SPI IrDA/ASK CAN/LIN CC-Link

Optional measurement boards along with a variety of measurement
cables expand the application range of the LE Series.

▲New communications standards different in ▲Cables and terminal adapters in a wide variety
hardware specification are supported with theare available to meet the shapes of thereplacement of the measurement board.
connectors of measurement targets.
>> see page 12

TTL/I 2 C/SPI Communications Expansion Kit OP-SB5GL
OP-SB5GL is an expansion kit for communication of RS-232C(V.24) and TTL/CMOS signal
level (1.8V - 5.0V). The measurement pod processes the signal, and the pod is connected withthe analyzer by the relay cable. Thus it has 1 meter or more reach for a prove unit to the target
than the TTL measurement port of LE-3500R/LE-2500R. Also it supports send/receive clock of
USART communication which is not supported by TTL measurement port of
LE-3500R/LE-2500R.

SDA
SCL

* I2C (Inter-Integrated Circuit) is a serialinterface standard for peripheraldevices developed by Philips
Corporation.

[I2C protocol setting screen]

RS-232C, TTL / CMOS (for I 2C and SPI)SD (SDA/SDO) , RD (SDI) , RS (SS) , CS , EX IN , SD CLK (SCL/SCK),
RD CLK , Trigger IN , Trigger OUT [Lead length: 170mm]
I 2C, SPI, BURST
Monitor / Simulation / BERT (*1 )It depends on the communication speed of the analyzer or the expansion firmware.
For the simulation of I2C, the rate is 50K, 100K, 200K, 384K, 417K or 1Mbps.

Interface
Probe signal
ProtocolTest function
Baud Rate[Example of I2C monitor display]

TTL/CMOS signal level

5.0V/3.3V/2.5V/1.8V signal levels can be selected.
When 5.0V selected - High: minimum 3.5V, Low: max 1.5V
When 3.3V selected - High: minimum 2.0V, Low: max 0.8V
When 2.5V selected - High: minimum 1.7V, Low: max 0.7V
When 1.8V selected - High: minimum 1.2V, Low: max 0.6V
Dedicated expansion board, relay cable, probe pod, prove unit

Input Level
Composition*1: BURST is available only for monitoring. I2C / SPI do not support BERT.

RS-530 Expansion Board OP-SB10N
OP-SB10N is an expansion board to equip LE-3500R / LE-2500R witha RS-530 port (Synchronous balanced communication such as X.21,
RS-449, V.35). The RS-422/485 interface of LE-3500R/LE-2500R is
terminal of 5 poles and does not have control lines for RS-422 while
OP-SB10R has the control lines of RTS/CTS and DSR/DTR. With
OP-SB10N, you can monitor/simulate the communication which needs
these control lines.

External signal
input/output

RS-232C(V.24)
measurement port

Interface
Test function
Composition

RS-422/485 (RS-530)
measurement port

RS-232C(V.24), RS-422/485 (RS-530)
Monitor, Simulation, BERT
Expansion board

To use the following cables for LE-3500R/LE-2500R, OP-SB10N is needed. These cables
are to be connected with the RS-422/485 (RS-530) connector of OP-SB10N.
>> see page12
Name

Model

Remarks

X.21 monitor cableLE-25Y15

Y shape branch cable, DSUB 15pin for X.20/21

RS-449 monitor cableLE-25Y37

Y shape branch cable, DSUB 37pin for RS-449

V.35 monitor cableLE-25M34

Y shape branch cable, M shape 34pin for V.35

RS-530 cableLE-25S530

Straight cable (shielded)

Current Loop Communication Expansion Kit OP-SB1C
OP-SB1C is an expansion kit for current loop communications which is presently used in the FA
field and resistant to external noise. You can monitor the communication of current loop up to
60mA. You can also output communication data of current loop with passive and active(20mA/40mA).

Transmission: Active device
Reception: Passive device

OP-1C

Current direction
SO

Note : OP-1C, an adapter for current loop included in this kit can be purchased separately. If you have
OP-SB5GL, with OP-1C you can use them as the same function that of OP-SB1C.

SI

Current direction

+V

R

RI

RO

* The current loop communications system is noise-resistant communications method
in which logic is decided with the ON/OFF of the current.

Interface

RS-232C, Current loop communications (4-pole terminal block)

Measurement signal

SD, RD

Baud rate (bps)19.2 kbps max. (*1)

Function

Monitor/Simulation

Monitor current level

10〜60mA

Signal polarity

Normal/inversion (*2)

Simulation mode

Passive test and active test
Active current: 20/40 mA (selectable with DIP switch)

OP-1C dimensions and mass

60(W)×100(D)×20(H)mm, Approx. 180g

OP-1C accessories

Dedicated expension board, current loop adapter (OP-1C),
relay cable [length : 800mm]*1: The baud rate is restricted by the cable length and current value.
*2: The signal polarity is set in the analyzer. (DIP switches for polarity switching equivalent to that of theconverntional model, OP-1B, are not provided.)Increases in the efficiency of developing and testing in-vehicle networks.
CAN/LIN Communications Expansion Kit OP-SB7GX
This expansion kit makes the measurement of up to 2 channels simultaneously byusing Controller Area Nerwork (CAN) communications used widely in FA systems andin-vehicle communications, and Local Interconnect Network (LIN) communications datain flexible connection. This expansion kit allows the simultaneous logic measurement
and analog measurement of four-line external signals.

●CAN/LIN Simultaneous Monitoring
The OP-SB7GX enables the simultaneous measurement of CAN
communications data and LIN communications data along with time stamp,
thus contributing to the development of bridge units connecting the CAN
and LIN. Furthermore, the ID filter can be used for highly efficient analysis.

LIN*CAN is the communications protocol for in-vehicle network
developed by Robert Bosch GmbH. LIN is communications protocol for
in-vehicle network proposed by automobile manufacturers in Europe.

CAN[Example of CAN/LIN monitor display]TM

Displays the time of frame reception completion in millisecond units (Example: 4216898 → Reception at 42 minutes 16.898 seconds)
The [ZOOM/CODE] key can be used to select the display of the difference in time (dT) from the moment the previous frame is received.

CH

Reception channels (1: CH1 and 2: CH2)ID

CAN: ID of received frame, LIN: Displays the following items in sequence. SynchBreak bit width, SynchFiled, "-" and ID(Example: 1355-35, SynchBreak width=13 bits, SynchField=55h, and ID=35h)

TYP

Types of reception frames

DAT: CAN data frame

REM: CAN remote frame

ERR: CAN error frame

FRM: LIN frame ILL: Illegal frames beyond LIN standards
CAN: Contents of data length code (number of data bytes) LIN: Data length set for CONFIG for each ID

DL
DATA

Contents of data fieldC

Contents of LIN checksum (hexadecimal)

S

Displays whether or not the frame was normal.

I

Displays the logic of external signal IN1. All the external signals will be displayed with the [DISPLAY MODE] key.

●CAN Simulation Function
A frame registered in the CAN data table is transmitted. A partof the data in the frame can be specified as sweep data thatcan be transmitted with the value of the data automaticallychanged from the initial value to the third target value, whichmakes it easy to check the response of the equipment
according to the change of communications data.

[Settings for CAN data table][Image of sweep data settings]
DATA

Sixteen frames specified with the number of repeating times
and transmission intervals can be registered in advance.

TIME●LIN Simulation Function
The OP-SB7GX in master mode can transmit the contents of
the LIN data table in the order set in the schedule tablerepeatedly or according to key manipulation. A parity error, any
number of break bits, and any SYNC data can be set to
conduct confirmation tests for error data with ease. While inslave mode, the contents of the data table set with an ID
conforming to the request of the master will be transmitted.
Furthermore, the WakeUp signal (80h) can be transmitted at
any time.

Applicable analyzer[Example of master mode settings]

A 15-bit response space (the space between the header
block and response block) and 5-bit inter-byte space (thespace between adjacent response data items) are set.

CAN: Conforms to ISO11898/ISO11519-2 standards (*1) (DSUB9-pin connectors x 2)

Transceiver

CAN: Equivalent to TJA1050/1054

No. of measurement channels

2 channels in total with CAN, LIN or CAN / LIN in combination

ProtocolCAN2.0B, device net, LIN (Rev 1.1, 1.2, 1.3, 2.0 and 2.1)

Monitor function
Trigger function
Simulation function
External signal input
Composition

Data table numbers 0, F, 1, and 2 are transmitted insequence with a parity error set for an 18-bit-length BREAK
filed for table F data.

LE-3500R, LE-2500R, LE-3500, LE-2500

Interface

Baud rate (bps)[Example of schedule table settings]

CAN: 1 Mbps max.

LIN: Conforms to ISO9141 standards (header 3-pin connector x 2)

LIN: Equivalent to TJA1021

LIN: 26 kbps max.

Arbitrary baud rate settings possible
ID filter possible and time stamp (1 ms min.) recordable
CAN: Standard/Expansion format supported and possible to make bit timing settings

LIN: Frame breaking possible according to the data length of each ID or specified idle time

Conditions: Specified communications data string (8 characters max.), specified remote frame (CAN), frame error (LIN), timer and counter coincidence, and external signal logic
Operation: Measurement stop, saving in a memory card, timer control, counter control, specified data transmission, buzzer, and validation of trigger conditions
Pre-registered data is selected by key operation and transmitted (sweep transmission available) and Master and slave simulation (LIN only).
Real-time display of 4-channel external signal state with LED possible.
Signal logic recordable in synchronization with data continuous measurement of voltage possible. (Measurement range: ±15 V)
Dedicated expansion board, DB9 monitor cable x 2, 3-wire probe cable x 2, and 8-wire probe cable*1: Settings in the Analyzer selectable with a relay.

High-speed HDLC/SPI Communications Firmware
This expansion firmware increases the baud rates of bit
synchronous communications (e.g., HDLC/SDLC/X.25,
and CC-Link communications)and SPI communications
up to 10 Mbps. The firmware processes main measurement
items completely with a field programmable gate array(FPGA), thus precisely capturing communications data
along with time stamps in 1-μs units. It is useful to measure
the high-speed HDLC communication at CC-LINK of
RS-485 multi-drop type, and the high-speed SPI/HDLC
communications at TTL signal level on the PCB boards.

OP-FW10R
Applicable analyzer

LE-3500R

Interface

RS-422/RS-485, TTL, SPI

ProtocolBaud rate (bps)Error check
Online monitor function

Simulation function[Screen for HDLC settings]

RS-485

PLC

HDLC, SDLC, X.25. CC-Link (NRZ/NRZI format, AR clock), SPI
Half-duplex
Full duplex

115.2 kbps Ѹ 10 Mbps
115.2 kbps Ѹ 5 Mbps

FCS error (CRC-ITU-T), Abort, Short frameTime stamp

6 digits (0 to 524287) Resolution: 1ms, 100 μ s, 10 μ s, 1 μ s units (selectable)

ID filter (HDLC)

2 characters can be set (Don’t care and Bit mask are available.)

Data table

16k data (can be divided into 16 parts and registered)

MANUAL mode

It transmits the registered data which corresponds with the key.
Repeat transmission and repeat interval can be set.

Trigger function

Max 8 characters (Don’t care and bit mask are available.). It automatically stops monitoring when it
detects two pairs of single or sequential, an error, or low level of the external signal trigger input.

Data search function

Search of trigger data, error data, and character line are available.

Composition

Firmware CD-ROM, manualLE-3500R / LE-2500R Specifications
LE-3500R

LE-2500R

RS-232C （V. 24）◎◎

RS-422/485 (*1)◎◎

TTL/I 2C/SPI◎◎

X. 20/21○ [ OP-SB10N ＋ LE-25Y15 ]○ [ OP-SB10N ＋ LE-25Y15 ]RS-449○ [ OP-SB10N ＋ LE-25Y37 ]○ [ OP-SB10N ＋ LE-25Y37 ]V. 35○ [ OP-SB10N ＋ LE-25M34 ]○ [ OP-SB10N ＋ LE-25M34 ] (*2)

TTL(USART)○ [ OP-SB5GL ]○ [ OP-SB5GL ]Current loop

○ [ OP-SB1C ]○ [ OP-SB1C ]CAN/LIN○ [ OP-SB7GX ]○ [ OP-SB7GX ]High-speed HDLC/CC-Link○ [ OP-FW10R ]−ASYNC (Asynchronous), ASYNC-PPP◎◎

Character synchronous SYNC/BSC◎◎

Bit synchronous HDLC/SDLC/X.25

◎

◎

I 2C

◎

◎

BURST (*3)◎◎

Modbus◎

Not supported

CC-Link◎

Not supported

CAN

◎

◎

Devicenet

◎

◎

LIN

◎

◎

Model
Interface

Expansion measurement
interface.

Expansion firmware

Standard Protocol

Optional Protocol

ST1 (DTE transmission clock), ST2 (DCE transmission clock), RT (DCE reception clock),
AR (The synchronous clock extracted from the edge of the transmission and reception data)

Synchronous clock
Capture memory

Baud rate

Memory capacity (*4)
Added function with memory used

64 MB
Two divided areas, data protection, and selection between fixed-size buffer and ring buffer, Auto save when finishing measurement.

Max. speed (full-duplex)

2.048Mbps

Max. speed (half-duplex)

2.048Mbps

1.000Mbps

50bps〜2.048Mbps

50bps〜1.000Mbps

Speed setting range
Speed setting step, accuracy

1.000Mbps

Freely set to four effective digits, separately for transmission and reception.(Margin of error: ±0.01% or less)

Data format

NRZ, NRZI, FM0, FM1

Data code

ASCII, EBCDIC, JIS7, JIS8, Baudot, Transcode, IPARS, EBCD, EBCDIK, HEX
Asynchronous

Character Framing

Character synchronous
Bit-oriented synchronous

Data bit (8 bits)

MP (multiprocessor) bit is shown with a special mark.

Multi-processor bit

LSB first or MSB first (switchable)

Bit transmission order

Normal or Inverted (switchable)

Polarity inversion
For all protocols
For bit-oriented synchronous protocol
Specification
Idle time display

Parity (ODD, EVEN, MARK, SPACE), Framing, Break, BCC (LRC, CRC-6, CRC-12, CRC-16, FCS-16, CRC-ITU-T, FCS-32).
BCC permeation mode.
Abort, short frameCommunication log is recorded continuously and displayed in the LCD without affecting the communication lines.
OFF (no recording); Resolusion: 100ms, 10ms, 1ms; Max 999. 9 sec

Time stamp display

Date time stamp: Unit selectable among "Day/Hr/Min," "Hr/Min/Sec," "Min/Sec/10ms," or OFF (no recording)Line status display

Records and displays the wave form of 4 signals (chosen from RS(RTS), CS(CTS), ER(DTR), DR(DSR), CD(DCD),
CI(RI), EXIN(external trigger input) along with the transmission/ reception data.

Online monitor functions
Address filter
Data display and operations
Bit shift display
Protocol translation display
Target signals
Line status LED

Data bit + parity bit (6 or 8 bits in total)

NONE, ODD, EVEN, MARK, SPACE

Parity bit

Error check

Data bit (5, 6, 7, 8) + parity bit (0, 1) + stop bit (1, 2)

RS-232C
Other interface

Records only frames of the specified address. (only when HDLC/SDLC/X.25)
Pause in capture, scroll, paging, jump to the specified screen.
Entire frame can be shifted to the right or left in 1 bit increments.
SDLC (modulo 8/128), ITU-T X.25 (modulo 8/128), LAPD, PPP, BSC, I2C
Two color LEDs of SD, RD, RS(RTS), CS(CTS), ER(DTR), DR(DSR), CD(DCD), CI(RI), ST1(TXC1), ST2(TXC2), RT(RXC).
Logic ON (red) , logic OFF (green) , no connection NC (light off)
Logic ON (red) , logic OFF or no connection NC (light off)
2kinds; Max. count: 999999 (Resolution: 1ms ,10ms ,100ms)

Interval timer

2kinds; Max. count: 999999

General-purpose counter

For SD and RD (1 each): Max. count: 4294967295

Data counter
Simultaneous detection conditions

Up to 4 pairs of trigger condition and action can be specified. Sequential actions, which validate another condition after one condition is satisfied, is also possible.

Trigger condition

Communication error (Parity, MP, framing, BCC, break, abort, short frame can be specified individually.),
communication data string up to 8 characters (don't care and bit mask available), idle time more than the specified
duration, match time/counter value, logic status of interface signal line and extarnal trigger input

Trigger action

Stops measurement/test (offset can be set), validates trigger condition, controls timer (start/stop/restart), controls counter(count/clear), activates buzzer, saves monitor data on a memory card, sends the specified character string (during manual
simulation), and sends pulse output to external trigger terminal OT2.

External trigger output

Sends pulse to external trigger terminal OT1 when all conditions are satisfied. Sends pulse to external trigger terminal OT2 according to the trigger output specification.

Trigger function

Specification
Data search function

Search condition
Search action

Retrieves the data with specific condition from capture memory.
Communication error (parity, MP, framing, BCC, break, abort, short frame), communication data string up to 8 characters (don't care andbit mask available), idle time more than the specified duration, specified timestamp, and trigger matching data.
Shows the match data at the top or enumeration display (selectable)

Monitor conditions auto setting

Measurement conditions such as protocol, transmission speed, (max. 115.2Kbps), data code, synchronous character and BCC check can be set(*5).

Auto run/stop function

Enables measurement to start and end at the specified time at the selected repeating cycle (monthly, daily, hourly).
Enables measurement to start automatically after power is turned ON.

Power ON auto run function
Specification
Auto save function

Automatically saves the monitored data in the capture memory and saves as communications log file in a SD card or USB flash.

File size

BUF (capture memory size), 1MB, 2MB, 4MB, 8MB, 16MB, 32MB

Max. files1024LE-3500R

Model

Measures and displays the value of voltage amplitude: SD, RD, ER (DTR), and CD (DCD) over RS-232C(current/min/max, range ±15V (RS-232C), ±5.5V (TTL)).

Signal voltage measuring function

Takes statistics and displays graphs of transmission/reception data
count, number of frames, and satisfied trigger condition count.

Statistical analysis function

Logic analyzer function

LE-2500R

Measures and displays the interval of change in the interface signal line. (current/min/max/average, resolution: 0.1ms)

Delay time function

Specification

Measures the logical change of the interface signal in the sampling clock period, and displays its wave.

Sampling clock

1KHz to 20MHz (14 steps)

Sampling memory

Min 2,000

Trigger condition

Trigger conditions in the ONLINE monitor functions match. Logical status of interface signal or external signal match.

Trigger position

Before, center, after
×8 , ×4 , ×2 , ×1 , ×1/2 , ×1/4 , ×1/8 , ×1/16 , ×1/32 , ×1/64

Zoom in/out
Other functions

Time measurement by cursor, signal line exchange, signal status searchSpecification

Conforming to ITU-T G.821 it measures line quality such bit error rate and block error rate(*6).

Communication mode

Synchronous (SYNC), Asynchronous (ASYNC) RTS/CTS flow control is available.

Measuring speed
BERT

Measurement mode(bit error rate test)

Test pattern

50bps~2.048Mbps, freely set to four effective digits

26-1, 29-1, 211-1, MARK, SPACE, ALT, DBL-ALT, 3in24, 1in16, 1in8, 1in4

Error bit insertion

Inserts 1-bit or 5-bit error in test pattern by key operation.

Measurement range

It is able to measure the parameter of the ITU-T advice G.821. It is able to output the external trigger by detecting the error bit.
Effective received bit (0〜9999999), bit errors (0 to 9999999 to 9.99E9), bit error rate(0 to 9.99E-9 to 1),
block errors (0 to 9999999 to 9.99E9), block error rate (0 to 9.99E-9 to 1), Savail(available measurement time: 0 to 9999999sec),loss count (synch loss: 0 to 9999), error duration (0 to 9999999sec), %EFS (normal operation rate: 0.000 to 100.000%)
Enables transmission/reception test of any given data in DTE or DCE mode (selectable).

Transmission data entry

A part of transmission data can be registerd as error data such as parity error.

Line control mode

Auto Controls transmission timing with RS(RTS), CS(CTS), ER(DTR), CD(DCD) signal lines automatically in 1 ms
increments or manual (key operation) can be selected.

MANUAL mode

(Manual test)

FLOW mode(Flow control test)

ECHO mode(Echo test)

POLLING mode

(Multi-polling test)

PROGRAM mode

Simulates the X-on /X-off control data and flow control procedures of RTS/CTS control line. (Sender and receiver selectable).(*7)
Sends the received data frame by frame (buffer echo), by data (character echo) or by loop back.
Simulates multi-polling communications procedures. (Slave and master selectable)(Program simulation)

File types
File operations
Max. capacity (*8)
Printout function

Auto control turning ON driver only during data transmission or manual mode linking with ER (DTR) or CD (DCD) key
operation can be selected during simuration of RS-485.
Sends the data assinged to operation keys each time a key is pressed, while checking communications status on the display. Can be used together with the trigger function.

(Buffer transmission test)

Specification
File management function

Register 160 kinds of transmission data tables (10 groups x 16, total 16K data)

Error data entry

Transmission driver control

BUFFER mode

50bps~1.000Mbps, freely set to four effective digits

Continuous measurememt, specifies the number of receiving bit, specifies the time to measure, repeatedly measurement at the unit of 1 - 1440minutes

Specification

Simulation function

Not available

Reproduces transmission of selected data (SD or RD)
captured in memory by monitor function.

Not available

Creates a simulation program (Max. type: 4, Max steps: 512)
using the dedicated commands (37 types) to test
the communication procedure.

Not available

Measurement data and condition can be saved in a SD card or USB flash. And the formatted data/condition can be used in the PC.
Measurement data (.DT), all measurement conditions (.SU), trigger save data (TG SAVEnn.DT), and auto save data (#nnnnnnn.DT)
Normal file display, file display by specified type/created date basis, save, load, delete, delete all, and format
32 GB

16 GB

Specified range of measurement data can be continuously printed in format corresponding to the display mode. Displayed images can be printed to make hard copies.

LCD
Remote control

Monochrome 240 x 64 dots with backlight
PC link software (light edition)(*9) is attached. The library to control the analyzer is available. Application for Android smart phone “ LE-REMOTE4 ” is available.

AUX(RS-232C) port

Mini DIN8 pin connector. Communication speed: 9600bps to 230.4Kbps (6 steps)
Print out data, Can be used with PC [LE-PC300R], Can be used to upgrade the firmware.

USB2.0 device port

B-connector in device side. Transfer data in High speed. Can be used with PC [LE-PC300R], Can be used to upgrade the firmware.

USB2.0 host port

A-connector on the host side for USB flash connection, it transfers data in High speed.

Wi-Fi connection (*10)

802.11 b/g/n for PC connection, used for LE-PC300R. STA mode (via an access point) and AP mode (directly with a PC) are selectable.

External power supply
Built-in battery
Temperature range

Provided AC adapter Input: 100 to 240 VAC at 50/60Hz
Nickel hydrogen battery (Model: P-19S), Battery operating time(*11) : About 7 hours, Battery Charging time: About 2.5hours
In operation : 0 to 40 degrees, In storage : -10 to 50 degrees

Humidity range

85% (RH) max.

StandardCE(class A), EMC(EN61326-1 : 2013)

Dimensions

210 ( W ) x 154 ( D ) x 38 ( H ) mm
About 760g

Mass
Accessories

Monitor cable for DSUB 25-pin (LE-25M1), Monitor cable for DSUB 9-pin (LE-009M1), DSUB25pin-9pin conversion adapter (LE-259AD),
5 wires TTL prove cable (LE-5LS), USB cable, AC adapter (6A-181WP09), carrying bag (LEB-01), Utility CD, instruction manual and warranty◎: Standard support. ○: Supported with option product in [ ].
*1: To monitor control line, expansion board “OP-SB10N” (which has RS-530 port) is needed. *2: The control signal lines of V.35 are not supported. *3: This mode is used to capture all data in synchronization withclock edges. *4: Transmission/reception data, idle time, time stamp, and line status items consume 4 bytes of memory at each capture. *5: This function supports only ASYNC, SYNC/BSC, HDLC/SDLC. Correct auto
settings are impossible if the amount of communications data is small or communications data includes a large number of errors. *6: Only ASYNC mode and SYNC mode are available. *7: Only ASYNC is available.
*8: Only the SD cards purchased from LINEEYE are supported. *9: This is the light edition of optional product “LE-PC300R” (with some limitation for its functions). *10: Wi-Fi function is available only in Japan, USA, andCanada. *11: The battery operating time was measured under LINEEYE's measurement conditions with the LCD backlight turned OFF.

Order Information
Standard Set● LE-3500R/LE-2500R ……………… (Comes with Japanese manual.)● LE-3500R-E/LE-2500R-E ………… (Comes with English manual.)● Portable communication analyzer ………………

1● DSUB 25-pin monitor cable (LE-25M1) ………… 1● DSUB 9-pin monitor cable (LE-009M1) ………… 1● Conversion adapter for DSUB 25-pin

to DSUB 9-pin (LE259-AD) ……………………… 1● TTL prove cable (LE-5LS) ………………………… 1● USB cable …………………………………………… 1● AC adapter (6A-181WP09) ………………………

1
1● Utility CD (the light edition of PC link software is included.) … 1● Instruction manual ………………………………… 1● Warranty …………………………………………… 1● Carrying bag (LEB-01) ……………………………

An easy-totransport carrying
bag is provided.

Options for LE-3500 LE-2500● Cables / Terminal blocks / Converter
Monitor cable for DSUB 25-pin

Monitor cable for DSUB 9-pin

Conversion adapter for DSUB 25-pin to DSUB 9-pin

LE-25M1

LE-259M1

LE-259AD

Branch cable for monitoring communicationlines over general DSUB 25-pin.

Branch cable for measuring RS-232C over
DSUB 9-pin of PC, etc.

It converts DSUB 25-pin of RS-232C toDSUB 9-pin. Using with LE-009M1 it isequivalent with LE-259M1.
DB25(male) - DB9(female)

1.5m

0.1m

DB25(Male)

1.5m

DB25(Male)

DB25(Male) DB25(Female)

0.2m*Same as the one included in the accessories
of LE-3500R/LE-2500R.

DB9(Female) DB9(Male)*Same as the cable packed with analyzer.

X.21 Monitor cable

RS-449 Monitor cable

5 wires TTL prove cableLE-25Y15

LE-25Y37

LE-5LS

Branch cable for measuring X.20/21 over
DSUB 15-pin. (Shield type)

Branch cable for measuring RS-449 over
DSUB 37-pin. (Shield type)

1.2m

1.2m

This is the prove cable which is compatible withthe 5-pins TTL measurement port and external
trigger terminal of LE-3500R/LE-2500R. The
cable length is 350mm.

DB15(Male)*Same as the one included in the accessories
of LE-3500R/LE-2500R.

DB37(Male)

DB25(Male)

DB25(Male)
DB15(Female)

DB37(Female)

V.35 Monitor cable

RS-530 cable

AUX cable for DSUB 9-pin

DSUB 9-pin monitor cableLE-25M34

LE-25S530

LE2-8V

LE-009M1

Branch cable for measuring
V.35 over M34-pin.

A twisted pair cable
RS-530(shield type)

Cable for connection AUX (RS-232C)
port of an analyzer with PC (DSUB
9-pin DTE specification).

This is the branch cable for DSUB
9-pin.

for*Same as the one included in theaccessories of LE-3500R/LE-2500R,
and OP-SB7GX.

• Length:2.5m

1.5m
1.5m

*Same as the cable packed with analyzer.

M34(Male)

1.5m

DB25(Male)

DB25(Male)

M34(Female)

0.1m

DB25(Male)
DB9(Male)● Memory card

DB9(Female) DB9(Male)● Carrying bag

32GB SDHC card

16GB SDHC card

SD-32GX

SD-16GX

LEB-01The SD-card which is confirmed to operate for
LINEEYE analyzer
Supported model: LE-3500R

The SD-card which is confirmed to operate for
LINEEYE analyzer
Supported model: LE-3500R, LE-2500R

Bag with pockets for storing and carrying
accessories such as AC adapter, cables, etc.

*The image is for illustration purposes.

*The image is for illustration purposes.

● AC Adapter

Carrying bag*Same as the carring bag packed with analyzer.

● Battery pack

Wide input AC adapter

NiMH battery pack for replacement

6A-181WP09

P-19S

Input: AC100-240V, 50/60Hz
Output: DC9V, 2A
Plug: center

Rating: 4.8V, 1900mAH
Applicable model:LE-3500R, LE-2500R, LE-3500,
LE-2500, LE-1500,LE-7200,
LE-3200,LE-2200, LE-1200*Same as the AC adapter packed with analyzer.

Compact thermal Printer RoHS

*An auxiliary and replacement battery equivalent to
the Analyzer built-in battery.

MULTI PROTOCOLANALYZER

Options
Directive
Compatible

Compact thermal printerCompact Thermal Printer Set

DPU-414-51B-E

DPU-414-PA

Includes printer (DPU-414-51B-E),
roll paper x1, AC adapter, and printer*AC adapter and cable are not prepared. cable (LE2-8P).
Built-in battery, dedicated roll paper(x1) included.
Provide them separately.

Handy thermal printer for on-site
printout of measurements• Prints 40 digits per line in normal mode and 80 digits in reduced mode.
• High-speed printing at 52.5 characters per second.
• Incorporates eco-friendly NiMH battery.
• Supports Centronics parallel and RS-232C ports.
• Dimensions: 160(W)x 170(D)x 67(H)mm• Weight: Approx. 690g (including built-in NiMH battery)AC adapter for DPU-414

PW-C0725-W2
Input: AC100Vࠥ240V
Output: DC7.0V, 2.5A(center)

Roll paper

TP-411L
Thermal roll paper for DPU-414.
10 rolls per carton.
Width: 112mm Length per roll: Approx. 28m

LE-8200A/
LE-8200

240（W）×190（D）×48（H）mm,about1.1kg

Battery pack for DPU-414

BP-4005-E
Same as NiMH battery built-in
DPU-414-51B-E.
4.8V, 1100mAh

The top-level model of battery-powered portable communications analyzer with wide color display.
●Measurement at Low to Mega Speed up ●A long recording time of communications logs
to 4Mbps.
on 128GB CF card or USB flash drive (*1).
●Logic analyzer function and analog waveform analysis (*2).
●100 Mbytes capture memory.
●A compact and lightweight model in B5 size
●Supports to TTL, I2C, SPI, CAN, LIN,
FlexRay, LAN and USB.
operating continuously for 4 hours.

AUX cable for DPU-414

LE2-8P
This is a cable to connect AUX(RS-232C) port of the analyzer withthe serial port of DPU-414.
The length is 1.5m.

Read the instruction manual provided with the product before use and use the productexplained in that manual. Using the product in ways not guaranteed in the manual,
SAFETY asconnecting
it to systems outside of the specified ranges and remodeling can all cause
and damage. LINEEYE CO. LTD. will assume no responsibility whatsoever for
WARNING trouble
trouble or damage arising because of unauthorized ways of use.

SisterProduct*1 USB flash is supported only by LE-8200A. *2 High-speed analog waveform analysis requires an optional device.

• All brand names and product names mentioned in this catalog are trademarks or registered trademarks of theirrespective companies.
• Specifications and designs of products listed in this catalog are as of June 2017, and are subject to change
without notice for improvement.
• Colors of actual products may differ slightly from that listed due to printing condition.
• This catalog may not be reprinted or duplicated, in part or in whole.
©2017 by LINEEYE CO., LTD.

LINEEYE CO., LTD.
Head Office/Sales Office
Marufuku Bldg 4F, 39-1 Karahashi Nishihiragaki-cho, Minami-ku, Kyoto, 601-8468
PHONE: 81-75-693-0161 FAX:81-75-693-0163* LINEEYE CO. LTD. is a venture company founded by electronicequipment development members of the former Sekisui Chemical Co.,
Ltd. with investment from the Sekisui Venture Fund. The electronicequipment business of Sekisui Electronic Co. Ltd. was transferred to
LINEEYE CO. LTD. in October 2000.

L-17601E/LE G
Printed in Japan