R&S®RT-Zxx
HIGH BANDWIDTH PROBES
Specifications

Data Sheet | Version 25.00

Version 25.00, November 2019

CONTENTS
Definitions.............................................................................................................................................................................................. 3
Probe/oscilloscope chart ........................................................................................................................................................................ 4
R&S®RT-ZZ80 transmission line probe .................................................................................................................................................. 6
R&S®RT-ZS10L active probe ................................................................................................................................................................. 8
R&S®RT-ZS10/-ZS10E/-ZS20/-ZS30 active probes ............................................................................................................................. 11
R&S®RT-ZS60 active probe ................................................................................................................................................................. 16
R&S®RT-ZD10/-ZD20/-ZD30 differential probes .................................................................................................................................. 19
R&S®RT-ZD40 differential probe.......................................................................................................................................................... 24
R&S®RT-ZM15/-ZM30/-ZM60/-ZM90/-ZM130/-ZM160 modular probes ............................................................................................... 27
R&S®RT-ZPR20/-ZPR40 power rail probes ......................................................................................................................................... 34
Ordering information ............................................................................................................................................................................ 37

2

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

Version 25.00, November 2019

Definitions
GeneralProduct data applies under the following conditions:Three hours storage at ambient temperature followed by 30 minutes warm-up operation
Specified environmental conditions met
Recommended calibration interval adhered toSpecifications with limits
Represent warranted product performance by means of a range of values for the specified parameter. These specifications are marked
with limiting symbols such as <, ≤, >, ≥, ±, or descriptions such as maximum, limit of, minimum. Compliance is ensured by testing or isderived from the design. Test limits are narrowed by guard bands to take into account measurement uncertainties, drift and aging, if
applicable.
Specifications without limits
Represent warranted product performance for the specified parameter. These specifications are not specially marked and represent
values with no or negligible deviations from the given value (e.g. dimensions or resolution of a setting parameter). Compliance is ensured
by design.
Typical data (typ.)Characterizes product performance by means of representative information for the given parameter. When marked with <, > or as arange, it represents the performance met by approximately 80 % of the instruments at production time. Otherwise, it represents the meanvalue.
Measured values (meas.)
Characterize expected product performance by means of measurement results gained from individual samples.
Typical data as well as measured values are not warranted by Rohde & Schwarz.

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

3Version 25.00, November 2019

1

4

RT-ZA9

Page

1 GHz

2 GHz

4 GHz

6 GHz

4 GHz

6 GHz

8 GHz

13 GHz16 GHz

○

●

●

●

●

●

●

●

●

●

●

●

○

6

○
●
●
○
○
○

○
○
○
●
○
○

○
●
●
○
○
○

○
○
○
●
○
○

○
○
○
○
●
○

○
○
○
○
●
●

○
○
○
○
●
●

○
○
○
○
●
●

○
○
○
○
●
●

○
○
○
○
●
●

○
○
○
○
●
●

○
○
○
○
●
●

●
●
●
●
●
●

8
11
11
11
11
16

●
○
○
○

○
●
○
○

●
○
○
○

○
●
○
○

○
○
●
○

○
○
○
●

○
○
○
●

○
○
○
●

○
○
○
●

○
○
○
●

○
○
○
●

○
○
○
●

●
●
●
●

19
19
19
24

RTP

RTE
RTO
≤ 600 MHz

Passive probes
RT-ZZ80
Active probes
RT-ZS10L 1
RT-ZS10E
RT-ZS10
RT-ZS20
RT-ZS30
RT-ZS60
Differential probes
RT-ZD10
RT-ZD20
RT-ZD30
RT-ZD40

1 GHz

Probe:
R&S®

≤ 500 MHz

Base unit:
R&S®

RTM3000
RTA4000

Probe/oscilloscope chart

The R&S®RT-ZS10L has a BNC interface and requires 50 Ω input coupling. It can be attached to oscilloscopes with 1 MΩ input coupling using a
BNC feedthrough termination adapter.

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

●○

RT-ZA9

Page

1 GHz

2 GHz

4 GHz

6 GHz

4 GHz

6 GHz

8 GHz

13 GHz16 GHz

○
○
○
○
○
○

○
○
○
○
○
○

●
○
○
○
○
○

●
○
○
○
○
○

○
●
○
○
○
○

○
○
●
○
○
○

○
○
●
●
○
○

○
○
●
○
○
○

○
○
●
●
○
○

○
○
○
●
○
○

○
○
○
○
●
○

○
○
○
○
○
●

○
○
○
○
○
○

27
27
27
27
27
27

●
○

●
○

●
○

●
○

●
○

○
●

○
●

○
●

○
●

○
●

○
●

○
●

●
●

34
34

RTP

RTE
RTO
≤ 600 MHz

Modular probes
RT-ZM15
RT-ZM30
RT-ZM60
RT-ZM90
RT-ZM130
RT-ZM160
Power rail probes
RT-ZPR20
RT-ZPR40

1 GHz

Probe:
R&S®

≤ 500 MHz

Base unit:
R&S®

RTM3000
RTA4000

Version 25.00, November 2019

recommended extra
possible accessory, with limited functionality of probe or base unit

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

5Version 25.00, November 2019

R&S®RT-ZZ80 transmission line probe
All parameters are valid for the probe only when connected to a host instrument with an input impedance of 50 Ω.
See table on page 4 and Rohde & Schwarz oscilloscope operating manual for more details.
R&S®RT-ZZ80
Step response
Rise time
10 % to 90 %Frequency response
Bandwidth
starting at DC
Input impedance
DC input resistancesystemInput capacitance
DC characteristics
Attenuation
systemAttenuation error
probe only, with ideal 50 Ω load impedance
Maximum nondestructive input voltage
Continuous voltage
ESD tolerance
human body model

6

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes< 60 ps
8.0 GHz (meas.)
500 Ω ± 1 %0.3 pF (meas.)
10:1
±1 %
20 V (RMS)
2 kV (meas.)

Version 25.00, November 2019

General data
Temperature
Temperature loading

operating temperature range
storage temperature range

Climatic loading

AltitudeCalibration interval
SafetyRoHS
Mechanical data
Dimensions

WeightProbe interface
Connector

operationtransport

probe head (L × W × H)
cable length
overall length
probe only

0 °C to +50 °C−40 °C to +70 °C+25 °C/+40 °C cyclic at 95 % relative
humidity without condensation,in line with IEC 60068-2-30
up to 3000 m
up to 4600 m
2 years
in line with IEC/EN 61010-1
in line with EN 50581
approx. 68 mm × 12 mm × 7.5 mm(2.68 in × 0.47 in × 0.3 in)approx. 1.1 m (43 in)approx. 1.2 m (48 in)approx. 40 g (0.1 lb)
SMA

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

7Version 25.00, November 2019

R&S®RT-ZS10L active probe
All parameters are valid when the probe is connected to an appropriate Rohde & Schwarz oscilloscope with an input impedance of 50 Ω.
See table on page 4 and Rohde & Schwarz oscilloscope operating manual for more details.
R&S®RT-ZS10L
Step response
Rise time
Frequency response
Bandwidth
Input impedance
DC input resistance
Input capacitance
DC characteristics
Dynamic range
Attenuation
Attenuation error
Maximum rated input voltage
DC
AC
Base unit
Input coupling8

10 % to 90 %, calculated from 0.35/bandwidth

350 psstarting at DC> 1 GHz (meas.)
1 MΩ (meas.)
0.9 pF (meas.)

derated, see figure on page 9probe only, with ideal 50 Ω load impedance

±8 V
10:1
±0.5 % (meas.)

derated, see figure on page 9±20 V
20 V (Vp)

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

50 Ω

Version 25.00, November 2019

R&S®RT-ZS10L maximum rated sine-wave peak voltage versus frequency

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

9Version 25.00, November 2019

General data
R&S®RT-ZS10L
Temperature
Temperature loading

operating temperature range
storage temperature range

Climatic loading

Altitude
SafetyRoHS
EMC
Calibration interval
Mechanical data
Dimensions

WeightProbe interface
Connector
Supply type10

operationtransport

probe head (L × W × H)
length of probe cableprobe only

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

0 °C to +50 °C−40 °C to +70 °C
80 % relative humidity for temperatures
up to +31 °C,
decreasing linearly to 40 % at +50 °C
up to 2000 m
up to 15 000 m
in line with EN 61010-1
in line with EN 50581
in line with EN 61326-1
2 years
approx. 111 mm × 22 mm × 14 mm(4.3 in × 0.9 in × 0.6 in)approx. 1.3 m (51 in)approx. 96 g (0.21 lb)
BNC
use the provided power adaptor only

Version 25.00, November 2019

R&S®RT-ZS10/-ZS10E/-ZS20/-ZS30 active probes
All parameters are valid for the probe only when connected to a host instrument with an input impedance of 50 Ω.
See table on page 4 and Rohde & Schwarz oscilloscope operating manual for more details.

Step response
Rise time
Flatness
Propagation delay
Frequency response
Bandwidth
Flatness

Input impedance
DC input resistance
Input capacitance
DC characteristics
Attenuation
Attenuation error
Temperature drift,
attenuation
Zero error

Temperature drift,
zero error

R&S®RT-ZS10/
R&S®RT-ZS10E

R&S®RT-ZS20

R&S®RT-ZS30

10 % to 90 %starting 2 ns after edge

< 350 ps
2 % (meas.)
5.5 ns (meas.)< 250 ps

< 135 ps

starting at DC, calculated from rise time
100 kHz to 100 MHz
100 MHz to 500 MHz
500 MHz to 1 GHz

1.0 GHz0.2 dB (meas.)––

1.5 GHz0.2 dB (meas.)
0.5 dB (meas.)–

3.0 GHz0.2 dB (meas.)
0.5 dB (meas.)
0.5 dB (meas.)1 MΩ
0.8 pF (meas.)

after applying digital correction factors

10:1±0.5 %±60 ppm/°C

after applying digital correction factors, referenced to probe input+15 °C to +35 °C±2 mV
0 °C to +50 °C±4 mV
referenced to probe input±90 µV/°C

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

11Version 25.00, November 2019

Dynamic range
DC
Offset compensation
range
Offset compensation error
AC
Total harmonic distortion

Noise voltage

not available with R&S®RT-ZS10E
offset compensation setting = 0 V
offset compensation setting ≠ 0 V
with zero or compensated DC component
16 V (Vpp) sine-wave input at
 300 MHz for R&S®RT-ZS10/-ZS10E
 1 GHz for R&S®RT-ZS20/-ZS30
referenced to probe input

Maximum nondestructive input voltage
DC peak voltage
AC peak voltage
derated, see figure
ESD tolerance
human body model

12

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes±8 V (+ offset compensation setting)±12 V
no additional error±0.2 % of setting ± 2 mV (meas.)
16 V (Vpp)–35 dB (meas.)

1.2 mV (RMS)(meas.)±30 V
30 V
8 kV (meas.)

1.6 mV (RMS)(meas.)

2.0 mV (RMS)(meas.)

Version 25.00, November 2019

Sine-wave peak voltage [V]

35
30
25
20
15
10
5
0
1E6

10E6

100E6

1E9

10E9

Frequency [Hz]Maximum nondestructive sine-wave peak voltage versus frequency

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

13Version 25.00, November 2019

R&S®ProbeMeter
Specifications for measurement error apply only when offset compensation setting is 0 V. The R&S ®RT-ZS10E probe is not equipped with
an R&S®ProbeMeter.
Measurement error+15 °C to +35 °C
0 °C to +50 °C±0.1 % of reading ± 750 µV±0.2 % of reading ± 1.5 mV±50 ppm/°C of reading ± 40 µV/°C> 87 dB
147 msoperating temperature range
storage temperature range

0 °C to +50 °C−40 °C to +70 °C+25 °C/+40 °C cyclic at 95 % relative
humidity without condensation,in line with IEC 60068-2-30
up to 3000 m
up to 4600 m

Temperature drift
50/60 Hz rejection
Integration time

General data
Temperature
Temperature loading
Climatic loading

Altitude
Mechanical resistanceVibration

operationtransport
sinusoidal

randomShock

14

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

5 Hz to 150 Hz, max. 2 g at 55 Hz,
0.5 g from 55 Hz to 150 Hz,
in line with EN 60068-2-6
10 Hz to 500 Hz,
acceleration 1.9 g (RMS),
in line with EN 60068-2-64
40 g shock spectrum,in line with MIL-STD-810E

Version 25.00, November 2019

EMC

Calibration interval
SafetyRoHS
Mechanical data
Dimensions

WeightProbe interface
Connector

in line with EMC Directive 2014/30/EC,
IEC/EN 61326-1 (table 2),
IEC/EN 61326-2-1,
CISPR 11/EN 55011(class B)
2 years
in line with IEC/EN 61010-1
in line with EN 50581
probe head (W × H × L)
cable length
overall length
probe only

approx. 12 mm × 7.5 mm × 68 mm(0.47 in × 0.3 in × 2.68 in)approx. 1.1 m (43 in)approx. 1.3 m (51 in)approx. 90 g (0.2 lb)
Rohde & Schwarz probe interface

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

15Version 25.00, November 2019

R&S®RT-ZS60 active probe
All parameters are valid for the probe only when connected to a host instrument with an input impedance of 50 Ω.
See table on page 4 and Rohde & Schwarz oscilloscope operating manual for more details.
R&S®RT-ZS60
Step response
Rise time
Flatness
Propagation delay
Frequency response
Bandwidth
Flatness
Input impedance
DC input resistance
Input capacitance
DC characteristics
Attenuation
Attenuation error
Temperature drift, attenuation
Zero error

Temperature drift, zero error

16

10 % to 90 %starting 2 ns after edge

< 67 ps
2 % (meas.)
5.5 ns (meas.)starting at DC, calculated from rise time
100 kHz to 100 MHz
100 MHz to 1 GHz

6.0 GHz0.3 dB (meas.)
0.3 dB (meas.)

see figure on page 18 for input impedance

1 MΩ
0.3 pF (meas.)
10:1

after applying digital correction factors
0 °C to +50 °C±0.5 %±100 ppm/°C
after applying digital correction factors, referenced to probe input+15 °C to +35 °C±2 mV
0 °C to +50 °C±4 mV
referenced to probe input±100 µV/°C

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

Version 25.00, November 2019

Dynamic range
DC
Offset compensation range
Offset compensation error
not when offset compensation setting = 0 V
ACwith zero or compensated DC component
Total harmonic distortion
16 V (Vpp) sine-wave input
Noise voltage
referenced to probe input
Maximum nondestructive input voltage
DC peak voltage
AC peak voltage
derated, see figure on page 18
ESD tolerance
human body model±8 V (+ offset compensation setting)±10 V±0.2 % of setting ± 2 mV (meas.)
16 V (Vpp)–70 dB (meas.)
2 mV (RMS) (meas.)
±30 V
30 V
2 kV (meas.)

R&S®ProbeMeter
Specifications for measurement error apply only when offset compensation setting is 0 V.
Measurement error
Temperature drift
50/60 Hz rejection
Integration time

+15 °C to +35 °C
0 °C to +50 °C±0.1 % of reading ± 2 mV
±0.2 % of reading ± 4 mV
±50 ppm/°C of reading ± 100 µV/°C> 87 dB
147 msGeneral data
See page 14.

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

17

35

10 M

30

1M
|Input impedance| [W]

Sine-wave peak voltage [V]Version 25.00, November 2019

25
20
15
10

10E6

100E6

1E9

10E9

Frequency [Hz]Maximum nondestructive sine-wave peak voltage versus
frequency

18

10 k
1k
100

5
0
1E6

100 k

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

10
100E0
100

1E3

10E3

100E3

1E6

10E6

100E6

Frequency [Hz]Input impedance versus frequency

1E9

10E9

Version 25.00, November 2019

R&S®RT-ZD10/-ZD20/-ZD30 differential probes
All parameters are valid for the probe only when connected to a host instrument with an input impedance of 50 Ω.
See table on page 4 and Rohde & Schwarz oscilloscope operating manual for more details.
Step response
Rise time
Flatness
Slew rate
Propagation delay
Frequency response
Bandwidth
Flatness

Common mode rejection

Input impedance
DC input resistance
Input capacitance

10 % to 90 %starting 2 ns after edge
referenced to probe input,
see figure on page 21

R&S®RT-ZD10

R&S®RT-ZD20

R&S®RT-ZD30< 350 ps
2 % (meas.)
60 V/ns (meas.)< 250 ps

< 135 ps

5.5 ns (meas.)starting at DC,
calculated from rise time
100 kHz to 100 MHz
100 MHz to 500 MHz
500 MHz to 1 GHz
DC to 10 kHz
10 kHz to 1 MHz
1 MHz to 1 GHz> 1 GHz

1.0 GHz

1.5 GHz

3.0 GHz0.2 dB (meas.)––> 50 dB
40 dB (meas.)
30 dB (meas.)
20 dB (meas.)

0.2 dB (meas.)
0.5 dB (meas.)–

0.2 dB (meas.)
0.5 dB (meas.)
0.5 dB (meas.)

between signal sockets
each signal socket to groundbetween signal sockets
each signal socket to ground1 MΩ
500 kΩ
0.6 pF (meas.)
0.8 pF (meas.)

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

19Version 25.00, November 2019

DC characteristics
Attenuation
Attenuation error
Temperature drift, attenuation
Zero error

after applying digital correction factors

Temperature drift, zero error
Dynamic range
Differential input
Offset compensation range
Offset compensation error

between signal sockets

Operating voltage window
CM offset compensation range
CM offset compensation error
Total harmonic distortion
Noise voltage

10:1±0.5 %±50 ppm/°C
after applying digital correction factors, referenced to probe input+15 °C to +35 °C±3 mV
0 °C to +50 °C±6 mV
referenced to probe input±150 µV/°C

offset compensation setting = 0 V
offset compensation setting ≠ 0 V
each signal socket to groundcm offset compensation setting = 0 V
cm offset compensation setting ≠ 0 V
10 V (Vpp) sine-wave input at 1 GHz
referenced to probe input

Maximum nondestructive input voltage
DC peak voltage
each signal socket to groundAC peak voltage
each signal socket to ground,derated, see figure on page 21
ESD tolerance
human body model,
each signal socket to ground2±5 V (+ offset compensation setting) 2±5 V
no additional error±0.2 % of setting ± 2 mV (meas.)±8 V (+ cm offset compensation setting) 2±22 V
no additional error±0.2 % of setting ± 2 mV (meas.)–35 dB (meas.)
1.7 mV (RMS)
2.4 mV (RMS)
3.0 mV (RMS)(meas.)(meas.)(meas.)±30 V
30 V
8 kV (meas.)

Can be extended with the R&S®RT-ZA15 external attenuator (see page 22). For the R&S®RT-ZD10, the R&S®RT-ZA15 is part of delivery.

20

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

Version 25.00, November 2019

Sine-wave peak voltage [V]

35
30
25
20
15
10
5
0
1E6

10E6

100E6

1E9

10E9

Frequency [Hz]Maximum nondestructive sine-wave peak voltage versus
frequency

Rise time versus input voltage (meas.)

R&S®ProbeMeter
Specifications for measurement error apply only when offset compensation setting is 0 V. The R&S ®ProbeMeter can be used to measuredifferential and common mode voltages.
Measurement error,
differential mode and common mode
Temperature drift
Common mode rejection
50/60 Hz rejection
Integration time

+15 °C to +35 °C
0 °C to +50 °C
for differential measurement,
0 °C to +50 °C±0.1 % of reading ± 2 mV
±0.2 % of reading ± 4 mV
±40 ppm/°C of reading ± 100 µV/°C> 50 dB> 87 dB
147 msRohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

21Version 25.00, November 2019

R&S®RT-ZA15 external attenuator
All parameters are valid for the R&S®RT-ZA15 external attenuator when connected to an R&S®RT-ZD10/-ZD20/-ZD30 differential probe.
R&S®RT-ZA15
Dynamic response
Rise time

Bandwidth

Common mode rejection

Input impedance
DC input resistance
Input capacitance
DC characteristics
System attenuation
External attenuation
External attenuation error
Dynamic range
Differential input
Offset compensation range
Operating voltage window
22

10 % to 90 %with R&S®RT-ZD10
with R&S®RT-ZD20
with R&S®RT-ZD30
starting at DC, calculated from rise timewith R&S®RT-ZD10
with R&S®RT-ZD20
with R&S®RT-ZD30
DC to 10 kHz, after adjustment
10 kHz to 1 MHz
1 MHz to 100 MHz
100 MHz to 1 GHz< 350 ps (meas.)< 250 ps (meas.)< 200 ps (meas.)
1.0 GHz
1.5 GHz
2.0 GHz
80 dB (meas.)
40 dB (meas.)
30 dB (meas.)
20 dB (meas.)

between signal sockets
each signal socket to groundbetween signal sockets
each signal socket to ground1 MΩ
500 kΩ
1.3 pF (meas.)
2.1 pF (meas.)probe and attenuator

100:1
10:1
±0.3 % (nom.)

between signal sockets±50 V (+ offset compensation setting)
±50 V
±60 V

each signal socket to groundRohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

Version 25.00, November 2019

Maximum rated input voltage
DC voltage
AC voltage
Transient peak voltage
ESD tolerance

each signal socket to groundeach signal socket to ground,derated, see figure
each signal socket to groundhuman body model,
each signal socket to ground±60 V
30 V (RMS)±42 V
8 kV (meas.)

Maximum rated sine-wave peak voltage versus frequency

General data
See page 14.

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

23Version 25.00, November 2019

R&S®RT-ZD40 differential probe
All parameters are valid for the probe only when connected to a host instrument with an input impedance of 50 Ω.
See table on page 4 and Rohde & Schwarz oscilloscope operating manual for more details.
R&S®RT-ZD40
Step response
Rise time
Flatness
Slew rate
Propagation delay
Frequency response
Bandwidth
Flatness

Common mode rejection

Input impedance
DC input resistance
Input capacitance24

10 % to 90 %starting 2 ns after edge
referenced to probe input, see figure on page 21< 90 ps, < 73 ps (meas.)
2 % (meas.)
60 V/ns (meas.)
5.5 ns (meas.)starting at DC, calculated from rise time
100 kHz to 100 MHz
100 MHz to 500 MHz
500 MHz to 1 GHz
DC to 10 kHz
10 kHz to 1 MHz
1 MHz to 1 GHz> 1 GHz

4.5 GHz, 5.5 GHz (meas.)
0.2 dB (meas.)
0.5 dB (meas.)
0.5 dB (meas.)> 50 dB
40 dB (meas.)
30 dB (meas.)
20 dB (meas.)

between signal sockets
each signal socket to groundbetween signal sockets
each signal socket to ground1 MΩ
500 kΩ
0.4 pF (meas.)0.65 pF (meas.)

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

Version 25.00, November 2019

DC characteristics
Attenuation
Attenuation error
Temperature drift, attenuation
Zero error

Temperature drift, zero error
Dynamic range
Differential input
Offset compensation range
Offset compensation error
Operating voltage window
CM offset compensation range 4
CM offset compensation error

10:1
after applying digital correction factors
0 °C to +50 °C±0.5 %±50 ppm/°C
after applying digital correction factors, referenced to probe input+15 °C to +35 °C±3 mV
0 °C to +50 °C±6 mV
referenced to probe input±150 µV/°C
between signal sockets
offset compensation setting = 0 V
offset compensation setting ≠ 0 V
each signal socket to groundcm offset compensation setting = 0 V
cm offset compensation setting ≠ 0 V
Total harmonic distortion
10 V (Vpp) sine-wave input at 1 GHz
Noise voltage
referenced to probe input
Maximum nondestructive input voltage
DC peak voltage
each signal socket to groundAC peak voltage
each signal socket to ground,see figure on page 21
ESD tolerance
human body model,
each signal socket to ground4±5 V (+ offset compensation setting)±5 V
no additional error±0.2 % of setting ± 2 mV (meas.)±8 V (+ cm offset compensation setting) 4±22 V
no additional error±0.2 % of setting ± 2 mV (meas.)–35 dB (meas.)
3 mV (RMS) (meas.)
±30 V
30 V
8 kV (meas.)

Available starting from serial number 200 000. Older probes have ±5 V operating voltage window and no cm offset compensation.

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

25Version 25.00, November 2019

R&S®ProbeMeter
See page 21.

General data
See page 14.

26

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

Version 25.00, November 2019

R&S®RT-ZM15/-ZM30/-ZM60/-ZM90/-ZM130/-ZM160 modular probes
All parameters are valid for the probe only when connected to a host instrument with an input impedance of 50 Ω. See table on page 4
and Rohde & Schwarz oscilloscope operating manual for more details.

R&S®RT-ZM15/-ZM30/-ZM60/-ZM90/-ZM130/-ZM160 modular probe amplifier
The R&S®RT-ZMxx modular probe amplifiers support four different measurement modes for differential mode (DM), common mode (CM)and single-ended measurements on the positive (P) and negative (N) signal terminal. The R&S®RT-ZMxx modular probe amplifiers
support two different attenuation settings (10:1 and 2:1).
Step response
Rise timeSlew rate
Propagation delay
Frequency response

10 % to 90 %, calculated from bandwidth
R&S®RT-ZM15
R&S®RT-ZM30
R&S®RT-ZM60
R&S®RT-ZM90
R&S®RT-ZM130
R&S®RT-ZM160
referenced to probe input, 10:1 attenuation
DM mode, 10:1 and 2:1 attenuation
R&S®RT-ZM15
R&S®RT-ZM30
R&S®RT-ZM60
R&S®RT-ZM90
R&S®RT-ZM130
R&S®RT-ZM160< 230 ps< 100 ps
< 75 ps< 50 ps< 35 ps< 28 ps80 V/ns (meas.)
5.5 ns (meas.)> 1.5 GHz> 3.0 GHz
> 6.0 GHz
> 9.0 GHz> 13.0 GHz> 16.0 GHz

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

27Version 25.00, November 2019

Common mode rejection

Input impedance
DC input resistance
DC characteristics
Attenuation
Attenuation error

Temperature drift, attenuation
Zero error
Temperature drift, zero error

28

DM mode, 10:1 and 2:1 attenuation
DC to 1 kHz
1 kHz to 1 MHz
1 MHz to 3 GHz
3 GHz to 6 GHz> 6 GHz> 50 dB
40 dB (meas.)
35 dB (meas.)
25 dB (meas.)
20 dB (meas.)

between signal terminals
each signal terminal to ground400 kΩ
200 kΩ

automatically set by oscilloscope vertical setting

10:1
2:1
±0.5 %

DM, CM, P, N mode, 10:1 and 2:1 attenuation,
after applying digital correction factors,
0 °C to +40 °C
after applying digital correction factors, referenced
to probe input
referenced to probe input

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes±40 ppm/°C (meas.)< 0.5 mV (meas.)±20 µV/°C (meas.)

Version 25.00, November 2019

Dynamic range
Input voltage range

DM, CM, P, N mode, 2:1 attenuation
DM, CM, P, N mode, 10:1 attenuation
Operating voltage window
each signal terminal to groundDM mode, 2:1 and 10:1 attenuation,
DC to 100 kHz
DM mode, 2:1 attenuation, > 100 kHz
DM mode, 10:1 attenuation, > 100 kHz
Offset compensation range
compensation of differential, common mode andsingle-ended offsets for each measurement mode
Offset compensation error
offset compensation setting = 0 V
offset compensation setting ≠ 0 V
Total harmonic distortion
DM mode, 5 V (Vpp) sine-wave input at 2 GHz
Noise voltage
DM mode, 2:1 attenuation,
for R&S®RT-ZM160 probe amplifier (16 GHz),
referenced to probe input
Noise spectral density
referenced to probe input
DM, CM, P, N mode, 2:1 attenuation
DM, CM, P, N mode, 10:1 attenuation
Maximum nondestructive input voltage
DC peak voltage
at probe tip module, each signal socket to groundAC peak voltage
at probe tip module, each signal socket to groundTermination voltage source
Compensation range
using VT socket at R&S®RT-ZMxx modular probe
amplifierOutput current
overcurrent turns termination voltage source off±0.5 V or 1.0 V (Vpp)±2.5 V or 5.0 V (Vpp)±7.0 V±1.0 V±5.0 V±16 V
no additional error±0.2 % of setting ± 2 mV (meas.)–31 dB (meas.)
5.4 mV (RMS) (meas.)

42 nV/rt(Hz) (meas.)
83 nV/rt(Hz) (meas.)
±30 V
±7 V
±4 V (meas.)±40 mA (meas.)

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

29Version 25.00, November 2019

R&S®ProbeMeter
Specifications for measurement error apply only when offset compensation setting is 0 V. The R&S®ProbeMeter simultaneously measuresdifferential, common mode, and single-ended DC components independent of the probe amplifier measurement mode setting.
Measurement error

0 °C to +40 °C

Common mode rejection

for differential measurement, 0 °C to +40 °C

30Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes±0.2 % of reading ± 5 mV
±0.05 % of reading ± 0.5 mV (meas.)> 50 dB

Version 25.00, November 2019

General data
Temperature
Temperature loading

operating temperature range
storage temperature range

Climatic loading

Altitude
Mechanical resistanceVibration

operationtransport
sinusoidal

randomShock
EMC

Calibration interval
SafetyRoHS
Mechanical data
Dimensions

WeightProbe interface

probe amplifier (W × H × L)
cable length
overall length
probe only

0 °C to +40 °C−40 °C to +70 °C+25 °C/+40 °C cyclic at 95 % relative humidity
without condensation,in line with IEC 60068-2-30
up to 3000 m
up to 4600 m
5 Hz to 150 Hz, max. 2 g at 55 Hz,
0.5 g from 55 Hz to 150 Hz,
in line with EN 60068-2-6
10 Hz to 500 Hz, acceleration 1.9 g (RMS),
in line with EN 60068-2-64
40 g shock spectrum,in line with MIL-STD-810E
in line with EMC Directive 2014/30/EC,
IEC/EN 61326-1 (table 2), IEC/EN 61326-2-1,
CISPR 11/EN 55011(class B)
2 years
in line with IEC/EN 61010-1
in line with EN 50581
approx. 18 mm × 10 mm × 105 mm(0.7 in × 0.4 in × 4.1 in)approx. 1.1 m (43 in)approx. 1.3 m (51 in)approx. 160 g (0.35 lb)
Rohde & Schwarz probe interface

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

31Version 25.00, November 2019

R&S®RT-ZMA probe tip modules
Frequency response
Bandwidth

Step response
Rise time

Input impedance
Input capacitance32

with R&S®RT-ZM160 probe amplifier (16 GHz)
R&S®RT-ZMA10 tip cable solder-in
R&S®RT-ZMA14 tip cable flex connect
R&S®RT-ZMA30 browser module
R&S®RT-ZMA40 SMA module
R&S®RT-ZMA50 extreme temperature kit
R&S®RT-ZMA12 tip cable square pin
R&S®RT-ZMA15 tip cable quick connect

16 GHz (meas.)

12 GHz (meas.)
6 GHz (meas.)
12 GHz (meas.)with R&S®RT-ZM160 probe amplifier (16 GHz),
calculated from bandwidth
R&S®RT-ZMA10 tip cable solder-in
R&S®RT-ZMA14 tip cable flex connect
R&S®RT-ZMA30 browser module
R&S®RT-ZMA40 SMA module
R&S®RT-ZMA50 extreme temperature kit
R&S®RT-ZMA12 tip cable square pin
R&S®RT-ZMA15 tip cable quick connect

10 % to 90 %with R&S®RT-ZMxx probe amplifier
R&S®RT-ZMA10 tip cable solder-in
R&S®RT-ZMA12 tip cable square pin
R&S®RT-ZMA14 tip cable flex connect
R&S®RT-ZMA15 tip cable quick connect
R&S®RT-ZMA30 browser module
R&S®RT-ZMA50 extreme temperature kit

differential
77 fF (meas.)279 fF (meas.)
90 fF (meas.)
109 fF (meas.)
32 fF (meas.)
77 fF (meas.)

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

28 ps

37 ps
75 ps
37 ps
single-ended
96 fF (meas.)521 fF (meas.)
144 fF (meas.)
150 fF (meas.)
52 fF (meas.)
96 fF (meas.)

Version 25.00, November 2019

DC input resistance

Input resistance > 3 MHz

Input return loss

Temperature
Temperature loading

RoHS

with R&S®RT-ZMxx probe amplifier
R&S®RT-ZMA10 tip cable solder-in
R&S®RT-ZMA12 tip cable square pin
R&S®RT-ZMA14 tip cable flex connect
R&S®RT-ZMA15 tip cable quick connect
R&S®RT-ZMA50 extreme temperature kit
R&S®RT-ZMA30 browser module
R&S®RT-ZMA40 SMA modulewith R&S®RT-ZMxx probe amplifier
R&S®RT-ZMA10 tip cable solder-in
R&S®RT-ZMA12 tip cable square pin
R&S®RT-ZMA14 tip cable flex connect
R&S®RT-ZMA15 tip cable quick connect
R&S®RT-ZMA50 extreme temperature kit
R&S®RT-ZMA30 browser module
R&S®RT-ZMA40 SMA module
R&S®RT-ZMA40 SMA module
3 kHz to 9 GHz
9 kHz to 16 GHz
operating temperature range
R&S®RT-ZMA10 tip cable solder-in
R&S®RT-ZMA12 tip cable square pin
R&S®RT-ZMA14 tip cable flex connect
R&S®RT-ZMA15 tip cable quick connect
R&S®RT-ZMA30 browser module
R&S®RT-ZMA40 SMA module
R&S®RT-ZMA50 extreme temperature kit

differential
400 kΩ

single-ended
200 kΩ

100 Ω
differential
764 Ω

60 Ω to VT
single-ended
382 Ω

100 Ω

50 Ω (> 3 kHz)> 12 dB (meas.)> 9 dB (meas.)> 12 dB (meas.)> 9 dB (meas.)–30 °C to +80 °C–30 °C to +80 °C–30 °C to +80 °C–30 °C to +80 °C
0 °C to +40 °C
0 °C to +40 °C–55 °C to +125 °C
in line with EN 50581

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

33Version 25.00, November 2019

R&S®RT-ZPR20/-ZPR40 power rail probes
All parameters are valid for the probe only with the supplied SMA extension cable when connected to a host instrument with an input
impedance of 50 Ω. See table on page 4 and Rohde & Schwarz oscilloscope operating manual for more details.
Step response
Rise time

Propagation delay
Frequency response
Bandwidth

Input impedance
Input resistance
Coupling capacitance
DC characteristics
Attenuation
Attenuation error
Temperature drift, attenuation
Zero error
Temperature drift, zero error

34

10 % to 90 %, calculated from bandwidth
R&S®RT-ZPR20
R&S®RT-ZPR40
R&S®RT-ZPR20
R&S®RT-ZPR40< 175 ps
< 120 ps, < 105 ps (meas.)
5.5 ns (meas.)
5.0 ns (meas.)starting at DC
R&S®RT-ZPR20
R&S®RT-ZPR40

2.0 GHz
3.5 GHz, 4.0 GHz (meas.)DCACin series with 50 Ω

50 kΩ
50 Ω
2 nF

after applying digital correction factors
after applying digital correction factors,
referenced to probe input
referenced to probe input

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

1:1±0.5 %±100 ppm/°C (meas.)±3 mV±75 µV/°C

Version 25.00, November 2019

Dynamic range
DC
Offset compensation range
Offset compensation error

offset compensation setting = 0 V
offset compensation setting ≠ 0 V
ACwith zero or compensated DC component
System noise voltage
referenced to probe input
Maximum nondestructive input voltage
DC peak voltage
AC peak voltage
ESD tolerance
human body model±850 mV (+ offset compensation setting)±60 V
no additional error±0.2 % of setting ± 1 mV (meas.)
1.7 V (Vpp)
10 % in addition to oscilloscope noise
±60 V
±5 V
8 kV (meas.)

R&S®ProbeMeter
Specifications for measurement error apply only when offset compensation setting is 0 V.
Measurement error

0 °C to +40 °C±0.2 % of reading ± 5 mV
±0.05 % of reading ± 0.1 mV (meas.)

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

35Version 25.00, November 2019

General data
Temperature
Temperature loading

operating temperature range
storage temperature range

Climatic loading

Altitude
Mechanical resistanceVibration

operationtransport
sinusoidal

randomShock
EMC

Calibration interval
SafetyRoHS
Mechanical data
Dimensions
WeightProbe interface
36probe box (W × H × L)probe box only

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

0 °C to +40 °C−40 °C to +70 °C+25 °C/+40 °C cyclic at 95 % relative
humidity without condensation,in line with IEC 60068-2-30
up to 3000 m
up to 4600 m
5 Hz to 150 Hz, max. 2 g at 55 Hz,
0.5 g from 55 Hz to 150 Hz,
in line with EN 60068-2-6
10 Hz to 500 Hz,
acceleration 1.9 g (RMS),
in line with EN 60068-2-64
40 g shock spectrum,in line with MIL-STD-810E
in line with EMC Directive 2014/30/EC,
IEC/EN 61326-1 (table 2),
IEC/EN 61326-2-1,
CISPR 11/EN 55011(class B)
2 years
in line with IEC/EN 61010-1
in line with EN 50581
approx. 29 mm × 41 mm × 96 mm(1.14 in × 1.61 in × 3.78 in)approx. 90 g (0.2 lb)
Rohde & Schwarz probe interface

Version 25.00, November 2019

Ordering information
Designation
Passive probes
8.0 GHz transmission line probe, 10:1, 500 Ω, 0.3 pF, 20 V (RMS)
Incl. signal pin, solder-in (50); ground pin, solder-in (10); signal pin (2); ground pin, pogo (2);signal adapter, square pin (2); ground adapter, square pin (2); SMA(f) to BNC(m) adapter;marker band kit; accessory box; carrying case; operating manual
Active probes
1.0 GHz active voltage probe, single-ended, 1 MΩ, 0.9 pF, BNC
Incl. spring tip 0.5 mm; solid tip 0.5 mm; marker band kit; ground blade; ground pin;
L-adapter; Y-lead adapter; ground lead, 6 cm; ground lead, 12 cm; ground lead, angled,
5 cm; ground lead, angled, 10 cm; micro clip (2); protective cap; power adapter; operating
manual
1.0 GHz active voltage probe, single-ended, 1 MΩ, 0.8 pF
Incl. R&S®RT-ZA2 accessory set; R&S®ProbeMeter; micro button
1.0 GHz active voltage probe, single-ended, 1 MΩ, 0.8 pF
Incl. signal pin (5); ground pin, pogo (2); ground pin, solderable, offset (2); marker band kit;mini clip (1); lead 15 cm (5.9 in) (1)1.5 GHz active voltage probe, single-ended, 1 MΩ, 0.8 pF
Incl. R&S®RT-ZA2 accessory set; R&S®ProbeMeter; micro button
3.0 GHz active voltage probe, single-ended, 1 MΩ, 0.8 pF
Incl. R&S®RT-ZA2 accessory set; R&S®ProbeMeter; micro button
6.0 GHz active voltage probe, single-ended, 1 MΩ, 0.3 pF
Incl. R&S®ProbeMeter; micro button
Incl. signal pin, solder-in (100); ground pin, solder-in (20); signal pin (5); ground pin,pogo (5); signal adapter, square pin (2); ground adapter, square pin (2); marker band kit;mini clip (2); micro clip (2); lead 6 cm (2.4 in) (2); lead 15 cm (5.9 in) (2); accessory box;
carrying case; operating manualType

Order No.

R&S®RT-ZZ80

1409.7608.02

R&S®RT-ZS10L

1333.0815.02

R&S®RT-ZS10

1410.4080.02

R&S®RT-ZS10E

1418.7007.02

R&S®RT-ZS20

1410.3502.02

R&S®RT-ZS30

1410.4309.02

R&S®RT-ZS60

1418.7307.02

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

37Version 25.00, November 2019

Designation
Differential probes
1.0 GHz active voltage probe, differential, 1 MΩ, 0.6 pF
Incl. R&S®ProbeMeter; micro button, R&S®RT-ZA15 external attenuator
See R&S®RT-ZD20 for additional equipment included
1.5 GHz active voltage probe, differential, 1 MΩ, 0.6 pF
Incl. R&S®ProbeMeter; micro button
Incl. signal pin, solder-in (10); signal pin, variable spacing (4); browser adapter; adapter,square pin (2); flex adapter, solder-in 4 cm (1.6 in) and 10 cm (3.9 in);flex adapter, square pin 4 cm (1.6 in) and 10 cm (3.9 in); lead 6 cm (2.4 in) (2);lead 15 cm (5.9 in) (1); mini clip (2); micro clip (2); marker band kit; carrying case;
operating manual
3.0 GHz active voltage probe, differential, 1 MΩ, 0.6 pF
See R&S®RT-ZD20 for equipment included
4.5 GHz active voltage probe, differential, 1 MΩ, 0.4 pF
Incl. R&S®ProbeMeter; micro button
Incl. signal pin, solder-in (100); socket adapter, variable spacing (2); browser adapter,
rigid (2); browser adapter, spring loaded (2); lead 6 cm (2.4 in) (2); lead 15 cm (5.9 in) (1);mini clip (2); micro clip (2); marker band kit; carrying case; operating manual
Power rail probes
2.0 GHz power rail probe, 1:1, low noise, 50 kΩ, large offset range ±60 V
Incl. R&S®ProbeMeter;
Incl. SMA extension cable, 1 m (39,4 in) (1), R&S®RT-ZA26 pigtail cable,
15 cm (5.9 in) (3); R&S®RT-ZA25 power rail browser kit (1); solder wire (1); adhesive pads
(7); marker band kit; carrying case; operating manual
4.0 GHz power rail probe, 1:1, low noise, 50 kΩ, large offset range ±60 V
Incl. R&S®ProbeMeter;
Incl. SMA extension cable low loss, 1.05 m (41.3 in) (1), R&S®RT-ZA26 pigtail cable,
15 cm (5.9 in) (3); R&S®RT-ZA25 power rail browser kit (1); solder wire (1); adhesive pads
(7); marker band kit; carrying case; operating manual38

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

Type

Order No.

R&S®RT-ZD10

1410.4715.02

R&S®RT-ZD20

1410.4409.02

R&S®RT-ZD30

1410.4609.02

R&S®RT-ZD40

1410.5205.02

R&S®RT-ZPR20

1800.5006.02

R&S®RT-ZPR40

1800.5406.02

Version 25.00, November 2019

Designation
Modular probes
1.5 GHz modular probe amplifier, differential, 400 kΩ, multimodeIncl. R&S®ProbeMeter; micro button
Incl. lead 15 cm (5.9 in) (1); Ag plated wire 0.1 mm (0.04 in) (1); Ag plated wire 0.2 mm(0.08 in) (1); solder wire (1); adhesive pads (7); solder-in resistor for R&S®RT-ZMA15 (20);solder lead for R&S®RT-ZMA12/15 (20); marker band kit; carrying case; operating manual
3.0 GHz modular probe amplifier, differential, 400 kΩ, multimodeIncl. R&S®ProbeMeter; micro button
See R&S®RT-ZM15 for additional equipment included
6.0 GHz modular probe amplifier, differential, 400 kΩ, multimodeIncl. R&S®ProbeMeter; micro button
See R&S®RT-ZM15 for additional equipment included
9.0 GHz modular probe amplifier, differential, 400 kΩ, multimodeIncl. R&S®ProbeMeter; micro button
See R&S®RT-ZM15 for additional equipment included
13.0 GHz modular probe amplifier, differential, 400 kΩ, multimodeIncl. R&S®ProbeMeter; micro button
See R&S®RT-ZM15 for additional equipment included
16.0 GHz modular probe amplifier, differential, 400 kΩ, multimodeIncl. R&S®ProbeMeter; micro button
See R&S®RT-ZM15 for additional equipment included
Storage case, for six R&S®RT-ZMA1x tip cables
Tip cable, solder in, length: 15 cm, multimode compatible
Tip cable, solder in, extended temperature, length: 15 cm, multimode compatible
Tip cable, square pin, for 1.27 mm pin header, length: 15 cm, multimode compatible
Tip cable, flex connect, solder-in, 15 cm, multimode compatible
Tip cable, quick connect, for solder in resistor connection, length: 15 cm,
multimode compatible

Type

Order No.

R&S®RT-ZM15

1800.4700.02

R&S®RT-ZM30

1419.3005.02

R&S®RT-ZM60

1419.3105.02

R&S®RT-ZM90

1419.3205.02

R&S®RT-ZM130

1800.4500.02

R&S®RT-ZM160

1800.4600.02

R&S®RT-ZMA1
R&S®RT-ZMA10
R&S®RT-ZMA11
R&S®RT-ZMA12
R&S®RT-ZMA14
R&S®RT-ZMA15

1409.3911.02
1419.4301.02
1419.4318.02
1419.4324.02
1338.1010.02
1419.4224.02

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

39Version 25.00, November 2019

Designation
Browser module, variable span from 0.5 mm to 8 mm, spring-loaded
Incl. spring loaded resistor tips (2 pairs)
SMA module, 2.92 mm/3.5 mm/SMA, differential, 100 Ω, DC termination,
multimode compatible; Incl. lead 11 cm (4.3 in) (1)Extended temperature kit, 1 m matched cable pair, multimode compatible
Incl. R&S®RT-ZMA11 (1); Ag plated wire 0.1 mm (0.04 in) (1); Ag plated wire 0.2 mm(0.08 in) (1); solder wire (1)Microwave probes
Differential microwave probe, 20 GHz, 0.3 mm pitch
Differential microwave probe, 20 GHz, 0.4 mm pitch
Differential microwave probe, 20 GHz, 0.5 mm pitch
Differential microwave probe, 20 GHz, 0.8 mm pitch
Differential microwave probe, 20 GHz, 1.0 mm pitch
Differential microwave probe, 20 GHz, 1.2 mm pitch
Single-ended microwave probe, 20 GHz, 0.25 mm pitch
Single-ended microwave probe, 20 GHz, 0.4 mm pitch
Single-ended microwave probe, 20 GHz, 0.5 mm pitch
Single-ended microwave probe, 18 GHz, 0.8 mm pitch
Single-ended microwave probe, 18 GHz, 1.0 mm pitch
Single-ended microwave probe, 16 GHz, 1.2 mm pitch
Single-ended microwave probe, 16 GHz, 1.4 mm pitch
Single-ended microwave probe, 16 GHz, 1.6 mm pitch
D probe handle for hand probing

40

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

Type
R&S®RT-ZMA30

Order No.
1419.4353.02

R&S®RT-ZMA40

1419.4201.02

R&S®RT-ZMA50

1419.4218.02

DPSS201503
DPSS201504
DPSS201505
DPSS201508
DPSS201510
DPSS201512
SPGR201502
SPGR201504
SPGR201505
SPGR181508
SPGR181510
SPGR161512
SPGR161514
SPGR161516
DP-HAND01

1801.4649.02
1801.4655.02
1801.4661.02
1801.4678.02
1801.4684.02
1801.4690.02
1801.4703.02
1801.4710.02
1801.4726.02
1801.4732.02
1801.4749.02
1801.4755.02
1801.4761.02
1801.4778.02
1801.4826.02

Version 25.00, November 2019

Designation
Accessories and sets
Spare accessory set for R&S®RT-ZS10/-ZS10E/-ZS20/-ZS30 active voltage probes
Contains: signal pin (10); ground pin, pogo (5); ground pin, solderable, offset (10); ground
adapter, square pin (2); marker band kit; mini clip (2); micro clip (2); lead 6 cm (2.4 in) (2);lead 15 cm (5.9 in) (2); accessory box; carrying case; operating manual
Pin set for R&S®RT-ZS10/-ZS10E/-ZS20/-ZS30 active voltage probes
Contains: signal pin (20); ground pin, pogo (5); ground pin, solderable, offset (20);ground adapter, square pin (2); marker band kit
Mini clips, contains: mini clip (10)Micro clips, contains: micro clip (4)Lead set, contains: lead 6 cm (2.4 in) (5); lead 15 cm (5.9 in) (5)Differential pin set for R&S®RT-ZD20/-ZD30
Contains: signal pin, solder-in (20); signal pin, variable spacing (10); browser adapter (2);adapter, square pin (2)Differential pin set for R&S®RT-ZD40
Contains: signal pin, solder-in (100); socket adapter, variable spacing (2); browser adapter,
rigid (2); browser adapter, spring loaded (2)Probe box to N/USB adapter
SMA(f) to BNC(m) adapter
External attenuator for R&S®RT-ZD10/-ZD20/-ZD30, incl. adjustment tool
Power rail browser kit
Contains: power rail browser, ground lead with alligator clip, ground spring (5), IC caps
(green, blue, gray, brown, black), insulating cap, tips (1 screwed on probe) (5), spring tip (5),sprung hook, micro SMD clip, dual adapter, flexible ground lead with adapter, coding rings(2 yellow, 2 red, 2 green, 2 blue), short tube (screwed on probe), long tube
Pigtail cable, solder-in, length: 15 cm, for R&S®RT-ZPR20
3D probe positioner
Power deskew fixture
Test fixture for probe characterization

Type

Order No.

R&S®RT-ZA2

1416.0405.02

R&S®RT-ZA3

1416.0411.02

R&S®RT-ZA4
R&S®RT-ZA5
R&S®RT-ZA6
R&S®RT-ZA7

1416.0428.02
1416.0434.02
1416.0440.02
1417.0609.02

R&S®RT-ZA8

1417.0867.02

R&S®RT-ZA9
R&S®RT-ZA10
R&S®RT-ZA15
R&S®RT-ZA25

1417.0909.02
1416.0457.02
1410.4744.02
1800.5329.00

R&S®RT-ZA26
R&S®RT-ZAP
R&S®RT-ZF20
R&S®RT-ZF30

1800.5258.00
1326.3641.02
1800.0004.02
1333.2099.02

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

41Version 25.00, November 2019

Service options
Extended warranty, one year
Extended warranty, two years
Extended warranty, three years
Extended warranty, four years
Extended warranty with calibration coverage, one year
Extended warranty with calibration coverage, two years
Extended warranty with calibration coverage, three years
Extended warranty with calibration coverage, four years
Extended warranty with accredited calibration coverage, one year
Extended warranty with accredited calibration coverage, two years
Extended warranty with accredited calibration coverage, three years
Extended warranty with accredited calibration coverage, four years

42

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

R&S®WE1
R&S®WE2
R&S®WE3
R&S®WE4
R&S®CW1
R&S®CW2
R&S®CW3
R&S®CW4
R&S®AW1
R&S®AW2
R&S®AW3
R&S®AW4

Please contact your local
Rohde & Schwarz sales
office.

Version 25.00, November 2019

Extended warranty with a term of one to four years (WE1 to WE4)
Repairs carried out during the contract term are free of charge 5. Necessary calibration and adjustments carried out during repairs arealso covered.
Extended warranty with calibration (CW1 to CW4)
Enhance your extended warranty by adding calibration coverage at a package price. This package ensures that your Rohde & Schwarz
product is regularly calibrated, inspected and maintained during the term of the contract. It includes all repairs 5 and calibration at therecommended intervals as well as any calibration carried out during repairs or option upgrades.
Extended warranty with accredited calibration (AW1 to AW4)
Enhance your extended warranty by adding accredited calibration coverage at a package price. This package ensures that yourRohde & Schwarz product is regularly calibrated under accreditation, inspected and maintained during the term of the contract. It includes
all repairs 5 and accredited calibration at the recommended intervals as well as any accredited calibration carried out during repairs or
option upgrades.

5

Excluding defects caused by incorrect operation or handling and force majeure. Wear-and-tear parts are not included.

Rohde & Schwarz R&S®RT-Zxx High Bandwidth Probes

43

Rohde & Schwarz GmbH & Co. KG
www.rohde-schwarz.com

Service that adds value

Regional contact► Worldwide► Local und personalized► Customized and ﬂexible► Uncompromising quality► Long-term dependability►► Europe, Africa, Middle East | +49 89 4129 123 45

customersupport@rohde-schwarz.com►► North America | 1 888 TEST RSA (1 888 837 87 72)

customer.support@rsa.rohde-schwarz.com►► Latin America | +1 410 910 79 88

customersupport.la@rohde-schwarz.com►► Asia Pacific | +65 65 13 04 88

customersupport.asia@rohde-schwarz.com►► China | +86 800 810 82 28 | +86 400 650 58 96

3607386822

customersupport.china@rohde-schwarz.com

Sustainable product design►► Environmental compatibility and eco-footprint►► Energy efficiency and low emissions►► Longevity and optimized total cost of ownership

ISO 9001

Certified Environmental Management

ISO 14001

R&S® is a registered trademark of Rohde & Schwarz GmbH & Co. KG
Trade names are trademarks of the owners
PD 3607.3868.22 | Version 25.00 | November 2019 (sk)
R&S®RT-Zxx High Bandwidth Probes
Data without tolerance limits is not binding | Subject to change© 2009 - 2019 Rohde & Schwarz GmbH & Co. KG | 81671 Munich, Germany

3607.3868.22 25.00 PDP 1 en

Certified Quality Management