HELiOSTM Series

User Manual (English)

HELiOS 300 & 850
User Controls & System Status Indicators
ISO 7000; Graphical symbols for use on equipment—Index and synopsis
Read user’s manual before operation.
Reg. # 1641
Storage or operating temperature
range. Reg. # 0632
Storage humidity range Reg. # 2620
Atmospheric pressure limitation. Reg.
# 2621
Keep away from rain, keep dry. Reg.
# 0626
Name and address of manufacturer.
Reg. # 3082
Caution, consult accompanying documents. Reg. # 0434A
Catalog Number. Reg. # 2493

Type BF applied part (degree ofprotection against electric shock). Reg.
# 5333Warning. Reg. # W001
Council Directive 93/42/EEC; concerning medical devices
Authorized representative in the European Community
If the product unique device identifier(UDI) label has the CE#### symbolon it, the device complies with therequirements of Directive 93/42/
EEC concerning medical devices. The
CE#### symbol indicates notified
body number.
This device complies with the requirements of Directive 2010/35/EU
concerning medical devices. It bears
the pi marking as shown.
ADR: European Agreement concerning theInternational Carriage of Dangerous Goods
by RoadNon-toxic gas.

Serial Number. Reg. # 2498
This way up. Reg. # 0623
Fragile, handle with care. Reg. #0621
ISO 7010: Graphical symbols—Safety colors
and safety signs—Registered safety signs
Frostbite may occur on contact withcold liquid or gaseous oxygen, or
frosted parts. Warning low temperature. To warn of low temperature or
freezing conditions. Reg. # W010

2 - ENG

Hazard Oxidizing substances: fire intensifying risk. Oxidizing agents cause
fires to burn more vigorously.
Refrigerated Liquid, USP; Produced
by Air Liquefaction
Internal Symbols
Keep unit well ventilated at all times
Keep away from flammable materials,
oil and grease.

Keep away from open flame, fire,
sparks. Open ignition source andsmoking prohibited. Reg. # P003

Portable Empty

Do not smoke near unit or while operating unit. Reg. # P002

Always keep Portable in one of thefollowing positions: upright, flat on its
back, or any position in between.

PN 21033264 C | User Manual

Portable Full

HELiOS 300 & 850
Wipe connector with clean dry cloth
before filling.
IEC 60417: Graphical Symbols for use on
equipment
Do not cover unit or carry portable
unit under your clothing. These units
normally vent oxygen. No. 5641
21 CFR 801.15: Code of Federal Regulations
Title 21
Federal law restricts this device to saleby or on the order of a physician.
Council Directive 2012/19/EU: waste electricaland electronic equipment (WEEE)
WEEE
IEC 60601-1: Medical electrical equipment Part
1 General requirements for basic safety and essential performance
Drip Proof
This product may be covered by one or more
patents, US and international. Please visit our
website below for the listing of applicable patents.
Pat.: www.caireinc.com/corporate/patents/

WARNING: THIS DEVICE IS NOT INTENDED
FOR LIFE SUSTAINING USE. IF CONTINUITY OF
OXYGEN SUPPLY IS REQUIRED, ENSURE THATAN ADEQUATE SUPPLY OF OXYGEN AND/OR A
SECONDARY OXYGEN SUPPLY IS AVAILABLE AT
ALL TIMES DURING THERAPY.
WARNING: IF YOU FEEL THE EQUIPMENT IS NOTOPERATING PROPERLY, CALL YOUR HEALTH CAREPROVIDER. DO NOT ATTEMPT TO REPAIR OR
ADJUST THE UNIT YOURSELF.
WARNING: DO NOT MODIFY THIS EQUIPMENTWITHOUT AUTHORIZATION FROM THE
MANUFACTURER.
WARNING: THE POSSIBILITY OF FIRE EXISTS
WHEN THE COMBINATION OF A FUEL, SOURCE
OF IGNITION, AND OXYGEN IS PRESENT.
HIGH CONCENTRATIONS OF OXYGEN (AIRIS APPROXIMATELY 21% OXYGEN) GREATLY
ENHANCE THE POSSIBILITY OF COMBUSTION.
WARNING: THE UNIT CONTAINS LIQUIDOXYGEN WHICH IS EXTREMELY COLD, ALMOST-300°F (-184 C°). EXPOSURE TO SUCH A LOWTEMPERATURE CAN CAUSE SEVERE FROSTBITE.
WARNING: LIQUID AND GASEOUS OXYGEN,THOUGH NONFLAMMABLE, CAUSE OTHER
MATERIALS TO BURN FASTER THAN NORMAL.
THIS HAZARD, ALONG WITH THE LOW
TEMPERATURE OF LIQUID OXYGEN, WARRANTS
CERTAIN SAFETY PRECAUTIONS.
WARNING: KEEP FLAMMABLE MATERIALSAWAY FROM THIS EQUIPMENT. AEROSOLSPRAYS, OILS AND GREASE, INCLUDING FACIAL
CREAMS AND PETROLEUM JELLY, IGNITE EASILYAND MAY BURN RAPIDLY IN THE PRESENCE OFOXYGEN.

Warning Information

WARNING: DO NOT USE OR STORE YOUR UNITWITHIN FIVE FEET OF ELECTRICAL APPLIANCES,ESPECIALLY STOVES, HEATERS, TOASTERS, AND
HAIR DRYERS.

Important: Read this manual thoroughly before
operating the H300/H850.
WARNING: A WARNING IDENTIFIESA SITUATION WHICH, IF NOTAVOIDED, COULD RESULT IN DEATH
OR SERIOUS INJURY. A WARNING
MAY ALSO DESCRIBE POTENTIALSERIOUS ADVERSE REACTIONS AND
SAFETY HAZARDS.

PN 21033264 C | User Manual

3 - ENG

HELiOS 300 & 850
WARNING: DO NOT SMOKE WHILEOPERATING THE UNIT.
WARNING: DO NOT ALLOW SMOKING, CANDLES, OR OPEN FLAMES IN
THE SAME ROOM WITH THE DEVICE.
WARNING: SMOKING WHILE WEARING ANOXYGEN CANNULA CAN CAUSE FACIAL BURNSAND POSSIBLY RESULT IN DEATH.
WARNING: REMOVING THE CANNULA AND
PLACING IT ON CLOTHING, BEDDING, SOFAS,OR OTHER CUSHION MATERIAL WILL CAUSE A
FLASH FIRE WHEN EXPOSED TO A CIGARETTE,HEAT SOURCE, SPARK OR FLAME.
WARNING: IF YOU SMOKE PLEASE: (1) TURN
OFF THE HELIOS PORTABLE, (2) TAKE OFF THE
CANNULA, AND (3) LEAVE THE ROOM WHERE
THE DEVICE IS LOCATED. WARNING: KEEP
YOUR UNIT IN A WELL-VENTILATED AREA. DO
NOT STORE LIQUID OXYGEN EQUIPMENT IN
A CLOSET, CAR TRUNK, OR OTHER CONFINED
AREA. DO NOT PLACE BLANKETS, DRAPERIES,OR OTHER FABRICS OVER EQUIPMENT. THIS WILLCAUSE AN OXYGEN ENRICHED ATMOSPHERE
WHICH IS A FIRE HAZARD.
WARNING: DO NOT CARRY THE PORTABLE UNIT
UNDER YOUR CLOTHING. DO NOT STORE LIQUIDOXYGEN EQUIPMENT IN A CLOSET, CAR TRUNK,OR OTHER CONFINED AREA. DO NOT PLACEBLANKETS, DRAPERIES, OR OTHER FABRICS
OVER EQUIPMENT. THIS WILL CAUSE AN OXYGENENRICHED ATMOSPHERE WHICH IS A FIRE
HAZARD.
WARNING: IN THE EVENT OF AN ACCIDENTAL
TIP-OVER, IMMEDIATELY BUT CAUTIOUSLY RETURN THE UNIT INTO AN UPRIGHT POSITION IF
POSSIBLE. IF ANY LIQUID OXYGEN IS ESCAPING,LEAVE THE AREA IMMEDIATELY AND CALL YOURHEALTHCARE PROVIDER. DO NOT ATTEMPT TO
MOVE THE UNIT OR STOP THE LIQUID OXYGENFROM ESCAPING.

WARNING: IN THE EVENT THERE IS A SERIOUS
INCIDENT OCCURRING WITH THIS DEVICE, THE
USER SHOULD IMMEDIATELY REPORT THE INCIDENT TO THE PROVIDER AND/OR THE MANUFACTURER. A SERIOUS INCIDENT IS DEFINED AS
AN INJURY, DEATH, OR POTENTIAL TO CAUSE
INJURY/DEATH SHOULD THERE BE A REOCCURRENCE OF THE INCIDENT. THE USER CAN ALSOREPORT THE INCIDENT TO THE COMPETENT AUTHORITY IN THE COUNTRY WHERE THE INCIDENTOCCURRED.
Caution: RX Only. Federal law restricts
this device to sale by or on the orderof a physician.
Caution: Use the H300/H850 Portable only as
directed by your doctor.
Caution: If the unit has been dropped, pleasecontact your health care provider to ensure theunit is still in working condition.
Caution: Do not allow untrained personnel tohandle or operate this device.
Caution: A caution identifies a potentially hazardous situation which, if not
avoided, may result in minor or moderate injury to the user or damage tothe equipment or other property.
Caution: RX Only. Federal law restricts thisdevice to sale by or on the order of a physician.
Caution: Use the H300/H850 Portable only as
directed by your doctor.
Caution: If the unit has been dropped, pleasecontact your health care provider to ensure theunit is still in working condition.
Caution: Do not allow untrained personnel tohandle or operate this device.

WARNING: DO NOT TOUCH FROSTED PARTS OFANY UNIT.

Note: Notes provide important information about
using the equipment properly.

WARNING: DO NOT STORE OR OPERATE THE HELIOS PORTABLE COUPLED TO YOUR RESERVOIR.

Note: Use of this device is prohibited on commercial
passenger and cargo air flights by the Federal Aviation Administration.

WARNING: THIS PRODUCT CAN EXPOSE YOU
TO CHEMICALS INCLUDING NICKEL, WHICH IS
KNOWN TO THE STATE OF CALIFORNIA TO CAUSECANCER. FOR MORE INFORMATION, GO TO
WWW.P65WARNINGS.CA.GOV.

4 - ENGPN 21033264 C | User Manual

Note: Applicable EMC information can be found in
the HELiOS Portable Technical Service Manual atwww.caireinc.com.

HELiOS 300 & 850
Specifications

Introduction• Mode of Operation: Demand and Continuous Flow
• Type of Protection Against Electrical Shock:Internally Powered Equipment• Degree of Protection Against Electrical
Shock: Type BF Applied Part
• IPX1 Classification According to the Degreeof Protection Against Ingress of Water: Drip
Proof
• Equipment not suitable for use in the presence of flammable mixtures

HELiOS Plus HELiOS(H300)
Marathon(H850)
LOX capacity

0.9 lb (0,41kg)

2.0 lb (0,91 kg)

Gaseousequivalentcapacity

308 L

685 L

Weight, filled

3.6 lb (1,63kg)

5.6 lb (2,54 kg)

Weight, empty

2.7 lb (1,22kg)

3.6 lb (1,63 kg)

Height including handle

10.5” (267mm)

15.0” (381 mm)

Typical use
time at 2 LPM
demand mode

9 hours

22 hours

Typical use
time at 2 LPM
continuous
mode

N/A

6.1 hours

Operatingpressure

22 psi (152kPa)

22 psi (152 kPa)

Normal evaporation rate

1.0 lb/day(0,45 kg/day)

0.85 lb/day(0,39 kg/day)Standard flow
control range

Continuous:
off, .12, .25, .5,.75 Demand:off, 1, 1.5, 2,
2.5, 3, 3.5, 4Continuous: off,
1,2,3,4,5,6
Demand: off,
1.5, 2, 2.5, 3, 4

A liquid oxygen system is designed to provide supplementary oxygen as prescribed by a physician.
Your liquid oxygen system includes a HELiOS
portable unit and a stationary reservoir. The reservoir is the larger container that is used to fill yourHELiOS portable. This user manual contains only
the instructions for using the HELiOS portable. Refer to the user manual supplied with your reservoir
unit for information on its operation.
HELiOS portable liquid oxygen units are designed
to be an ambulatory source of oxygen for an
extended period of time. They are filled with liquid
oxygen from a reservoir unit with a compatible fillconnector. HELiOS portables are available in two
different sizes.
The H300 Plus is primarily a conserving device
and holds 0.38 Liters of liquid oxygen. It provides
demand (pulse) flow settings 1-4, and limited low
dose continuous flow settings at 0.75 LPM andbelow only.
The H850 Marathon can be used as a conserving
or continuous flow device and holds 0.84 liters ofliquid oxygen. It provides continuous flow from 1-6
LPM and demand flow on settings of 1-4.

H300 and H850 portable
units shown.

Intended Use

The CAIRE HELiOS Liquid Oxygen unit is intended for the administration of supplemental oxygen.
The device is not intended for life support nor doesit provide any patient monitoring capabilities.

PN 21033264 C | User Manual

5 - ENG

HELiOS 300 & 850
Controls
1. Carrying Handle
2. Flow Control Knob
3. Dual Lumen Cannula Barbs
4. Contents Indicator Strap

5. Contents Indicator Window
6. Vent Valve Lever
7. Fill Connector (QDV)

1
2

4

3

5

7

Fill Connector Compatibility
The H300 and H850 can be filled from any CAIRE
top-fill (TF) reservoir that has a CAIRE-style malefill connector. The compatible male and female
connections for the HELiOS portable units areshown here.

CAIRE male (reservoir) and female (H300/H850) fill connectors
6 - ENGPN 21033264 C | User Manual6

HELiOS 300 & 850
Dual Lumen Cannula
In order to breathe from your H300 or H850, youmust do so through a dual lumen nasal cannula.
This type of cannula has two connections to theportable unit. One connection is used to detect
when you inhale, and the other connection is used
to deliver oxygen to your nose.
A single lumen cannula cannot be used on the
H300/H850. Your breath through the cannula provides the force necessary to open the flow control
valve and deliver oxygen. You must use a dual
lumen cannula and inhale through your nose on
ALL flow settings to ensure that oxygen is properly
delivered.
Note: Oxygen will only be delivered through the top
cannula barb.

Operating Instructions –Filling
1. Clean the fill connectors on the H300/H850 andthe reservoir with a clean, dry, lint-free cloth.
WARNING: THE FILL CONNECTORS MUST BE CLEAN AND DRY
WITH A LINT-FREE CLOTH ON BOTH
THE STATIONARY AND PORTABLEUNITS TO PREVENT FREEZING AND
POSSIBLE EQUIPMENT FAILURE.
WARNING: DO NOT DEPRESS ORDISTURB THE POPPET ON THE
FILL CONNECTOR WHEN DRYING
IT. THIS CAN CAUSE LEAKAGE OF
LIQUID OXYGEN.
2. Turn the flow control knob on the H300/H850 tothe off (“0”) position.
3. Hold the H300/H850 with both hands andposition its fill connector over the reservoir fillconnector.
4. Lower the portable onto the reservoir until you
feel the connectors engage.

Dual lumen nasal cannula operation: Oxygen is delivered throughtop port. Breath is detected through bottom port.

Note: To Equipment Provider: The following oxygen
administration accessories are recommended foruse with the HELiOS:• Nasal Cannula: CAIRE Part Number 6-778057-00• Firebreak: CAIRE Part Number 21126636
A firebreak is recommended for use with any
cannula.
• CAIRE offers a firebreak intended to be used inconjunction with the oxygen portable. The firebreakis a thermal fuse to stop the flow of gas in theevent that the downstream cannula or oxygen tubing is ignited and burns to the firebreak. It is placed
in-line with the nasal cannula or oxygen tubing
between the patient and the oxygen outlet of the
HELiOS. For proper use of the firebreak, always referto the manufacturer’s instructions (included witheach firebreak kit).
• For any additional recommended accessories,
please see the Accessories Catalog (PN MLLOX0010) available on www.caireinc.com.

5. Place one hand on top of the H300/H850 andapply a constant downward force.
6. Open the vent valve by pulling downward on thevent valve lever to begin the filling process. You
will hear a hissing sound that is perfectly normal.

PN 21033264 C | User Manual

7 - ENG

HELiOS 300 & 850
WARNING: DO NOT LEAVE UNIT
UNATTENDED WHILE FILLING.
WARNING: DO NOT OPERATE THE
H300/H850 WHILE ATTACHED TO
THE RESERVOIR UNIT.
WARNING: SHOULD LEAKAGE BE EXCESSIVE TO
THE POINT THAT A STREAM OF LIQUID IS PRESENT, LEAVE THE AREA AND CALL YOUR HEALTH
CARE PROVIDER IMMEDIATELY.

7. You may see vapor around the fill connection.
This is normal due to the great difference intemperature between the liquid oxygen and thewarm surrounding air.
8. Maintain a downward force on the H300/H850
with one hand during the entire filling process
and keep the vent valve lever open with the otherhand.
9. Approximately 20-30 seconds into the fill, closeand re-open the vent valve one or more times.
This will prevent the valve from freezing open.
10. The unit is full when the hissing sound changes
in tone and a dense white vapor is seen comingfrom under the shroud of the reservoir. When
this is observed, close the vent valve by raising
the vent valve lever. This will terminate the fill.
11. Hold the handle on the H300/H850 and push
the release button on the reservoir until the portable separates (release button varies by modelof reservoir).

WARNING: IF LARGE AMOUNTS OF VAPOR ARE
ISSUING FROM THE UNITS DURING FILLING,
STOP FILLING, LEAVE THE ROOM AND CALLYOUR HEALTH CARE PROVIDER.
WARNING: IF PROLONGED HISSING IS HEARD,
STOP USE AND CONTACT YOUR HEALTH CAREPROVIDER IMMEDIATELYWARNING: IF THE H300/H850 DOES NOT SEPARATE EASILY, DO NOT USE FORCE. THE UNITS
MAY BE FROZEN TOGETHER. LEAVE THE UNITSCONNECTED AND WAIT UNTIL THEY WARM UP– THEN THEY WILL SEPARATE EASILY. DO NOTTOUCH ANY FROSTED PARTS.
Caution: Should there be any
liquid leakage from the portable
after separating the units, , set
the H300/H850 aside, ensuring it
remains vertical, leave the room,and call your health care provider
immediately.
Caution: Should there be any liquid leakage
from the reservoir after separating the units,
open windows in the room, leave the room immediately, and call your health care provider.
Caution: Check the liquid level only after thevent valve is closed.
Caution: If the H300/H850 is being refilled
immediately after a period of use, fill it, andthen wait approximately one (1) hour after filling with the flow control knob set to off (“0”)
for the pressure to stabilize. Then operate as
normal.
NOTE: If the HELiOS portable is warm or has notbeen filled recently, it may take approximately 90
seconds to fill an H300 or approximately four (4)minutes to fill an H850.

12. Check the contents indicator to verify that theunit is full.
8 - ENGPN 21033264 C | User Manual

HELiOS 300 & 850
Operating Instructions –Breathing From the Portable
1. Attach both ends of a dual-lumen cannula to theconnection ports on the H300/H850. Either endof the cannula may be attached to either connection port on the portable.

NOTE: If the unit is on a demand (pulse) setting,
you will notice a small pulse at the beginning ofeach inhalation. Oxygen is delivered during thispulse, and continues as long as you inhale. Oxygen
flow stops when you stop inhaling.
NOTE: If the unit is in continuous flow, the H300/
H850 will deliver a steady stream of oxygen. You
must continue to breathe through the nasal cannula
normally to continue receiving continuous flow.
Note: Due to the possibility of excessive moisture/
ice build up it is recommended that the H300/
H850 remain idle for 1 hour after use to decreasemoisture buildup.
5. You should be receiving oxygen flow now.
Verify that you feel oxygen delivery.
6. Use the duration charts available at www.
CAIREmedical.com as a guideline to determine
the length of time the H300/H850 will operate.

2. Position the nasal cannula properly on your face
so that you will be able to breathe comfortably.
3. Turn the flow control knob until the setting prescribed by your physician is fully visible in thewindow on top of the H300/H850.

7. Under certain conditions (especially continuous
use when the portable is not moved) the H300/
H850 may develop frost on the case. You may
reduce this frost by wiping any accumulated
frost off the case.
8. To stop the flow of oxygen, turn the flow control
knob to the off (0) position.
Caution: Always turn the flow control knob off (0) when not in use.
NOTE: Do not open the vent valve during normal
operation.
NOTE: The H300/H850 may be operated in an
upright position, flat on its back, or any position in
between these two locations.

4. Breathe normally, inhaling through your nose.
Oxygen delivery will not begin until you inhale
through the nasal cannula.

PN 21033264 C | User Manual

9 - ENG

HELiOS 300 & 850
Operating Instructions –Level Verification
1. Lift the H300/H850 by the contents indicator
strap.
2. Push the bottom backside of the unit forward
until it is straight up and down.
3. Observe the green bar in the contents indicator window. The window is completely filledwith the green bar when the unit is full of liquid
oxygen.
Portable Full

Portable Empty

Checking the H300/H850 contents indicator

10 - ENG

PN 21033264 C | User Manual

User Troubleshooting
The following information is intended to help you
troubleshoot and solve simple operational problems
that you may experience when using your HELiOS
Portable.
1. The Portable makes a hissing sound.
• Hissing can occur to maintain the correct
operating pressure within the Portable. It is
most likely to hiss after filling or when theposition of the Portable is changed. Hissing
can last for approximately 10 minutes after
filling. Additionally, it can occur when theflow control valve is at a low setting.
2. The Portable does not pulse after filling.
• If during the filling process, the Portable isdisengaged with the vent valve lever down,pressure may be reduced within the Portable
causing a delay in the conserving device
function. It may require as much as 60 minutes to restore adequate pressure for accurate
oxygen flow. Additionally, improper fillingor lower than normal operating pressure in
the Reservoir will contribute to the unit notpulsing.
3. The Portable stops pulsing during use.
• Ensure that both connections of the dual
lumen cannula are firmly attached to the
Oxygen Outlet and Sensor connectors.
• Change the cannula if water droplets are present from humidified exhaled gas. Replacing
the cannula with a dry one enhances theability of the Portable to pulse.
• Ensure that the cannula is not kinked.
• Ensure that the cannula tips remain in yournostrils and do not slide to one side.
• Ensure that there is oxygen in the Portable.
• With the cannula on, close your mouth andbreathe only through your nose to verify that
the Portable has stopped pulsing.
4. The Portable does not fill.
• Verify that there is oxygen in the Reservoir.
• Ensure that the Portable and Reservoir fillconnectors are fully engaged throughout thefilling process.
5. The Portable vent valve does not close properlyat the end of the filling process.
• If the vent valve fails to close and the hissing
sound and oxygen vapor cloud continue,
carefully remove the Portable by depressing
the release button on the Reservoir. Venting
from the bottom of the Portable will stop ina few minutes. Allow the unit to warm until
you can close the vent valve. The Portable
may require as much as 60 minutes to restore

HELiOS 300 & 850
adequate pressure for accurate oxygen flow.
If needed, use an alternate source of oxygen
such as a flow control valve attached to theReservoir.
6. The Portable does not disengage easily from theReservoir after filling.
• The Portable and Reservoir fill connectors
may have become frozen. DO NOT USE
FORCE. Allow a few minutes for the frozen
parts to warm, then disengage the Portable
when the ice has melted. To prevent the units
from freezing together, always wipe the malefill connector on the Reservoir and the female
fill connector on the Portable with a clean,
dry cloth before filling.
NOTE: It is recommended that routine maintenancebe performed annually on this equipment. Pleasecontact your Health Care Provider to ensure thisroutine maintenance is being completed.

Cleaning Standard
WARNING: DO NOT ALLOW WATER INTO ANY OF THE CONTROLS,THE FILL CONNECTOR, OR THE
OXYGEN SUPPLY CONNECTOR.
Caution: Clean only after the unit
is empty and vented.

• Clean using a solution of mild dish washingdetergent and water.
• Apply cleaning solution directly to a lint-free
cloth. Do not spray cleaners directly on the
H300/H850.
• Wipe the outside surface with the lint-free
cloth until the outside surface is clean.

WEEE and RoHS
This symbol is to remind the equipment
owners to return it to a recycling facility
at the end of its life, per Waste Electricaland Electronic Equipment (WEEE)
Directive.
Our products will comply with the restriction ofHazardous Substances (RoHS) directive. They
will not contain more than trace amounts of lead orother hazardous material content.

Transport and Storage
The device should be stored in the upright position
well ventilated. Do not allow the device to be tilted
forward or laying on its front. Operating temperature range of 14°F to 104°F (-10°C to 40°C).
Storage temperature ranges from -40°F to 158°F(-40°C to 70°C).

Maintenance
Clean the fill connectors on both the stationary
and portable units with a clean, dry, lint-free cloth
between each fill to prevent freezing and possible
equipment failure.
NOTE: Any additional maintenance needed must
be done by a qualified service technician or service
provider.

Accessories

Caution: Do not use high temperature and high pressure washingequipment to clean these units.
• Do not get cleaner on any internal components or valves.
• Allow the unit to dry thoroughly before using.

Backpack­—HELiOS H850
Marathon

Hip pack­—HELiOS H300

H850 Cart

PN 21033264 C | User Manual

11 - ENG

w w w.ca ire inc . c om

CAIRE Inc.
2200 Airport Industrial Dr., Ste. 500
Ball Ground, GA 30107 U.S.A.
Medical Product Service GmbH
Borngasse 20
35619 Braunfels, Germany
Tel: +49 (0) 6442-962073
E-mail: info@mps-gmbh.eu
CAIRE and CAIRE Inc. are registered trademarks of CAIRE Inc. Please visit our website below for a
full listing of trademarks. Trademarks: www.caireinc.com/corporate/trademarks/.
Copyright © 2020 CAIRE Inc. CAIRE Inc. reserves the right to discontinue its products, orchange the prices, materials, equipment, quality, descriptions, specifications and/or, processes to
its products at any time without prior notice and with no further obligation or consequence. All
rights not expressly stated herein are reserved by us, as applicable.

|21033264-C<~

7 May 2020 21033264 C