User Manual

VisionAire 2 & 3

User Controls & System Status Indicators
SymbolDefinition

SymbolDefinition

Read user manual before operation.
See user manual for instructions.

ON (power switch on)

No Smoking Icon: Do not smoke nearunit.

OFF (power switch off)

Warnings and Cautions

No serviceable parts inside. Do not open
cover.

Type BF equipment (degree ofprotection against electric shock)Class II equipment

Complies with the 93/42/EEC directive
drawn up by the approved organization
No. 0459

Safety agency for CAN/CSA C22.2
No. 601.1 M90 for medical electricalequipment

Keep unit and accessories dry

Use no oil or grease.

Consult the accompanying documents

This symbol is to remind the equipment
owners to return it to a recycling facility
at the end of its life, per Waste Electricaland Electronic Equipment (WEEE)
Directive. Proper disposal of wasteof electrical and electronic equipment
required.

Oxygen outlet connection to thecannulaDo not expose to open flames

Caution: Federal law (USA) restricts
this for sale or rental by or on the orderof a physician or licensed health careprovider.

Keep in the vertical position

Fragile – handle with care

Oxygen concentration warning LED

Date of Manufacture

Name and address of manufacturer

Serial number

Authorized representative in theEuropean Community

Catalog number

2 — MN165-1 M

Contents
User Controls & System Status Indicators.................................................................................. 2
Indications for Use............................................................................................................................. 4
Contraindications for Use................................................................................................................ 4
Safety Guidelines................................................................................................................................ 4
Introduction......................................................................................................................................... 8
Concentrator Components............................................................................................................. 9Operating Instructions...................................................................................................................12
Care and Cleaning............................................................................................................................13
Troubleshooting...............................................................................................................................14
Product Specifications....................................................................................................................15
Disposal...............................................................................................................................................19

English: A multilingual version of the manual is available through your equipment provider.
Español: Una versión multilingüe del manual está disponible a través de su proveedor de equipo.
Français: Une version multilingue du manuel est disponible par l’intermédiaire de votre fournisseur dematériel.
Deutsche: Eine mehrsprachige Version des Handbuchs ist in Ihrer Geräte-Anbieter
VisionAire 2 & 3 User Manual — 3

Important!
Safety Instructions are defined as follows:
WARNING: IMPORTANT SAFETY INFORMATION
FOR HAZARDS THAT MIGHT CAUSE SERIOUS
INJURY.
CAUTION: Important information for preventing
damage to the VisionAire.
Note: Information needing special attention.

Indications for Use
WARNING: IT IS VERY IMPORTANT TO SELECTONLY THE PRESCRIBED LEVEL OF OXYGEN. DO
NOT CHANGE THE FLOW SELECTION UNLESSYOU HAVE BEEN DIRECTED TO DO SO BY A
LICENSED CLINICIAN.THE PORTABLE OXYGEN CONCENTRATOR MAY BE USED DURING SLEEP UNDER THE RECOMMENDATION OF A QUALIFIED CLINICIAN.
WARNING: FEDERAL (USA) LAW RESTRICTS THIS DEVICETO SALE OR RENTAL BY ORDER OF A PHYSICIAN OR OTHER
LICENSED HEALTH CARE PROVIDER.
WARNING: THIS UNIT IS NOT TO BE USED FOR LIFE
SUPPORT. GERIATRIC, PEDIATRIC, OR ANY OTHER PATIENTUNABLE TO COMMUNICATE DISCOMFORT WHILE USING
THIS DEVICE MAY REQUIRE ADDITIONAL MONITORING.
PATIENTS WITH HEARING AND/OR SIGHT IMPAIRMENT(S)
MAY NEED ASSISTANCE WITH MONITORING ALARMS.

Contraindications for Use
WARNING: IN CERTAIN CIRCUMSTANCES, THE
USE OF NON-PRESCRIBED OXYGEN CAN BE
HAZARDOUS. THIS DEVICE SHOULD ONLY BE
USED WHEN PRESCRIBED BY A PHYSICIAN.
WARNING: NOT FOR USE IN THE PRESENCE OF FLAMMABLE
ANESTHETICS.
WARNING: AS WITH ANY ELECTRICALLY POWERED DEVICE,THE USER MAY EXPERIENCE PERIODS OF NON-OPERATIONAS A RESULT OF ELECTRICAL POWER INTERRUPTION, OR
THE NEED TO HAVE THE VISIONAIRE SERVICED BY A QUALIFIED TECHNICIAN. THE VISIONAIRE IS NOT APPROPRIATEFOR ANY PATIENT WHO WOULD EXPERIENCE ADVERSEHEALTH CONSEQUENCES AS THE RESULT OF SUCH TEMPORARY INTERRUPTION.

Safety Guidelines
WARNING: NO MODIFICATION OF THIS EQUIPMENT IS PERMITTEDWARNING: SMOKING WHILE USING OXYGEN IS
THE NUMBER ONE CAUSE OF FIRE, INJURY, AND
DEATH. YOU MUST FOLLOW THESE SAFETYWARNINGS:WARNING: DO NOT ALLOW SMOKING, CANDLES, OR OPEN
FLAMES WITHIN THE SAME ROOM OF THE DEVICE OR THE
OXYGEN-CARRYING ACCESSORIES.
WARNING: SMOKING WHILE WEARING AN OXYGEN CANNULA CAN CAUSE FACIAL BURNS AND POSSIBLY RESULTIN DEATH.
WARNING: REMOVING THE CANNULA AND PLACING IT
ON CLOTHING, BEDDING, SOFAS, OR OTHER CUSHION
MATERIAL WILL CAUSE A FLASH FIRE WHEN EXPOSED TO A
CIGARETTE, HEAT SOURCE, SPARK OR OPEN FLAME.
WARNING: IF YOU SMOKE, YOU MUST ALWAYS FOLLOWTHESE 3 IMPORTANT STEPS FIRST: TURN OFF THE OXYGENCONCENTRATOR, TAKE OFF THE CANNULA, AND LEAVE THE
ROOM WHERE THIS DEVICE IS LOCATED.
CAUTION: Federal (USA) law restricts this device to sale or rental by order of a physician
or other licensed health care provider.
WARNING: THIS DEVICE SUPPLIES HIGH-CONCENTRATION OXYGEN THAT PROMOTES RAPID
BURNING. DO NOT ALLOW SMOKING OR OPEN
FLAMES WITHIN THE SAME ROOM OF (1) THIS
DEVICE, OR (2) ANY OXYGEN-CARRYING ACCESSORY. FAILURE TO OBSERVE THIS WARNING CAN RESULTIN SEVERE FIRE, PROPERTY DAMAGE AND / OR CAUSE
PHYSICAL INJURY OR DEATH.
WARNING: DO NOT USE YOUR OXYGEN CONCENTRATOR IN
THE PRESENCE OF FLAMMABLE GASES. THIS CAN RESULTIN RAPID BURNING CAUSING PROPERTY DAMAGE, BODILYINJURIES OR DEATH.
WARNING: DO NOT LEAVE A NASAL CANNULA ON CLOTHING, BED COVERINGS OR CHAIR CUSHIONS. IF THE UNIT IS
TURNED ON BUT NOT IN USE, THE OXYGEN WILL MAKE THE
MATERIAL FLAMMABLE. SET THE I/0 POWER SWITCH TO
THE 0 (OFF) POSITION WHEN THE OXYGEN CONCENTRATORIS NOT IN USE.
WARNING: USE NO OIL, GREASE, OR PETROLEUM-BASEDOR OTHER FLAMMABLE PRODUCTS WITH THE OXYGEN-CARRYING ACCESSORIES OR THE OXYGEN CONCENTRATOR. ONLY WATER BASED, OXYGEN COMPATIBLE
LOTIONS OR SALVES SHOULD BE USED. OXYGEN ACCELERATES THE COMBUSTION OF FLAMMABLE SUBSTANCES.
WARNING: CLEAN THE CABINET, CONTROL PANEL, AND
POWER CORD ONLY WITH A MILD HOUSEHOLD CLEANERAPPLIED WITH A DAMP CLOTH (NOT WET) OR SPONGE, AND
THEN WIPE ALL SURFACES DRY. DO NOT ALLOW ANY LIQUIDTO GET INSIDE THE DEVICE.
Note:PortableandmobileRFcommunicationsequipmentcan
effect medical electrical equipment.

4 — MN165-1 M

WARNING: DO NOT ATTEMPT ANY MAINTENANCE OTHER THAN THE POSSIBLE SOLUTIONS LISTED WITHIN THIS MANUAL. DO NOTREMOVE COVERS, ONLY YOUR EQUIPMENTPROVIDER OR A QUALIFIED SERVICETECHNICIAN SHOULDREMOVE THE COVERS OR SERVICE THIS DEVICE.
WARNING: ONLY ACCESSORIES RECOMMENDED BY
THE MANUFACTURER. USE OF ANY OTHER MAY BE
HAZARDOUS, CAUSE SERIOUS DAMAGETOYOUR OXYGENCONCENTRATOR AND WILL VOID THE WARRANTY .
WARNING: THE MANUFACTURER RECOMMENDS AN ALTERNATE SOURCE OF SUPPLEMENTAL OXYGEN IN THE EVENT OF A POWER
OUTAGE, ALARM CONDITION, OR MECHANICAL FAILURE. CONSULT YOUR PHYSICIAN OR EQUIPMENTPROVIDER FOR THE TYPE OF RESERVE SYSTEM REQUIRED.
WARNING: DO NOT ALLOW EITHER THE AIR INTAKE OR
THE AIR OUTLET VENTS TO BECOME BLOCKED. DO NOTDROP OR INSERT ANY OBJECTS INTO ANY OPENINGS ON
THE DEVICE. THIS CAN CAUSE THE OXYGEN CONCENTRATOR TO OVERHEAT AND IMPAIR PERFORMANCE.

CAUTION: In the event of an alarm or you
observe the Oxygen Concentrator is notworking properly; consult the Troubleshooting section in this manual. If you cannot
resolve the problem, consult your Equipment Provider.
CAUTION: To prevent a void warranty, follow all manufacturers’ instructions.
WARNING: ELECTRICAL SHOCK HAZARD.
TURN OFF THE UNIT AND DISCONNECT THE
POWER CORD FROM THE ELECTRIC OUTLETBEFORE YOU CLEAN THE UNIT TO PREVENT
ACCIDENTAL ELECTRICAL SHOCK AND BURN HAZARD.
ONLY YOUR EQUIPMENT PROVIDER OR A QUALIFIEDSERVICE TECHNICIAN SHOULD REMOVE THE COVERS ORSERVICE THE UNIT.
WARNING: CARE SHOULD BE TAKEN TO PREVENT THE
OXYGEN CONCENTRATOR FROM GETTING WET OR
ALLOWING FLUIDS TO ENTER THE UNIT. THIS CAN CAUSETHE UNIT TO MALFUNCTION OR SHUT DOWN, AND CAUSEAN INCREASED RISK FOR ELECTRICAL SHOCK OR BURNS.
WARNING: DO NOT USE LIQUID DIRECTLY ON THE UNIT.
A LIST OF UNDESIRABLE CHEMICAL AGENTS INCLUDES
BUT IS NOT LIMITED TO THE FOLLOWING: ALCOHOL
AND ALCOHOL-BASED PRODUCTS, CONCENTRATEDCHLORINE-BASEDPRODUCTS(ETHYLENECHLORIDE),AND
OIL-BASED PRODUCTS (PINE-SOL®, LESTOIL®). THESE
ARE NOT TO BE USED TO CLEAN THE PLASTIC HOUSING
ON THE OXYGEN CONCENTRATOR, AS THEY CAN DAMAGE
THE UNIT’S PLASTIC.

WARNING: SMOKING WHILE USING OXYGENIS THE NUMBER ONE CAUSE OF FIRE INJURIESAND RELATED DEATHS. YOU MUST FOLLOWTHESE SAFETY WARNINGS:WARNING: DO NOT ALLOW SMOKING, CANDLES, OR OPEN
FLAMES IN THE SAME ROOM WITH THE DEVICE OR THE
OXYGEN-CARRYING ACCESSORIES.
WARNING: SMOKING WHILE WEARING AN OXYGEN CANNULA MAY CAUSE FACIAL BURNS AND POSSIBLY DEATH.
WARNING: IF YOU SMOKE, THESE 3 STEPS MAY SAVE YOUR
LIFE: TURN OFF THE OXYGEN CONCENTRATOR, TAKE OFF
THE CANNULA, AND LEAVE THE ROOM WHERE THIS DEVICEIS LOCATED.
WARNING: “NO SMOKING – OXYGEN IN USE” SIGNS MUSTBE PROMINENTLY DISPLAYED IN THE HOME, OR WHERE OXYGEN IS IN USE. PATIENTS AND THEIR CAREGIVERS MUST
BE INFORMED ABOUT THE DANGERS OF SMOKING IN THE
PRESENCE OF, OR WHILE USING, MEDICAL OXYGEN.
WARNING: DO NOT USE YOUR OXYGEN CONCENTRATOR IN
THE PRESENCE OF FLAMMABLE GASES. THIS CAN RESULTIN RAPID BURNING CAUSING PROPERTY DAMAGE, BODILYINJURIES OR DEATH.
WARNING: REMOVING THE CANNULA AND PUTTING IT
ON CLOTHING, BEDDING, SOFAS, OR OTHER CUSHION
MATERIAL WILL CAUSE A FLASH FIRE WHEN EXPOSED TO A
CIGARETTE, HEAT SOURCE, OR FLAME.
WARNING: DO NOT LEAVE A NASAL CANNULA ON OR
UNDER CLOTHING, BED COVERINGS OR CHAIR CUSHIONS.
IF THE UNIT IS TURNED ON BUT NOT IN USE, THE OXYGENWILL MAKE THE MATERIAL FLAMMABLE. SET THE OXYGENCONCENTRATOR TO THE 0 (OFF) POSITION WHEN THE
OXYGEN CONCENTRATOR IS NOT IN USE.
CAUTION: Always place oxygen supply tubing
and power cords in a manner that prevents a
trip hazard.
WARNING: THE OXYGEN CONCENTRATORSHOULD NOT BE USED ADJACENT TO OR
STACKED WITH OTHER EQUIPMENT. IF ADJACENT OR STACKED USE IS UNAVOIDABLE,
THE DEVICE SHOULD BE OBSERVED TO VERIFY NORMAL
OPERATION.
WARNING: NO MODIFICATION OF THIS EQUIPMENT ISPERMITTED.
WARNING: USE OF CABLES AND ADAPTERS OTHER THAN
THOSE SPECIFIED, WITH THE EXCEPTION OF CABLES AND
ADAPTERS SOLD BY THE MANUFACTURER OF THE MEDICALELECTRICAL EQUIPMENT AS REPLACEMENT PARTS FOR
INTERNAL COMPONENTS, MAY RESULT IN INCREASEDEMISSIONS OF DECREASED IMMUNITY OF THE OXYGENCONCENTRATOR.
WARNING: USE ONLY ELECTRICAL VOLTAGE AS SPECIFIEDON THE SPECIFICATION LABEL AFFIXED TO THE DEVICE.
WARNING: DO NOT USE EXTENSION CORDS WITH THISUNIT OR CONNECT TOO MANY PLUGS INTO THE SAMEELECTRICAL OUTLET. THE USE OF EXTENSION CORDS
COULD ADVERSELY AFFECT THE PERFORMANCE OF THE
DEVICE. TOO MANY PLUGS INTO ONE OUTLET CAN RESULTIN AN OVERLOAD TO THE ELECTRICAL PANEL CAUSING THE
BREAKER/FUSE TO ACTIVATE OR FIRE IF THE BREAKER ORFUSE FAILS TO OPERATE.

VisionAire 2 & 3 User Manual — 5

AVERTISSEMENT: CET APPAREIL PRODUIT
DE L’OXYGÈNE À CONCENTRATION ÉLEVÉE,FAVORISANT UNE COMBUSTION RAPIDE. NE
PAS PERMETTRE DE FUMER OU DES FLAMMESNUES DANS LA MÊME CHAMBRE: (1) CET APPAREIL OU(2) TOUT ACCESSOIRE CONTENANT DE L’OXYGÈNE. NE
PAS UTILISER DE PRODUITS À BASE D’HUILE, DE GRAISSEOU DE PÉTROLE SUR OU À PROXIMITÉ DE L’UNITÉ.
DÉCONNECTER LE CORDON D’ALIMENTATION DE LA
PRISE ÉLECTRIQUE AVANT DE NETTOYER OU DE FAIRE
L’ENTRETIEN DE L’UNITÉ.
AVERTISSEMENT: RISQUE DE CHOC ÉLECTRIQUE. NE
PAS ENLEVER LES COUVERCLES LORSQUE L’UNITÉ ESTBRANCHÉE. SEULS VOTRE FOURNISSEUR D’ÉQUIPEMENTOU UN TECHNICIEN DE SERVICE QUALIFIÉ DEVRAITENLEVER LES COUVERCLES OU FAIRE L’ENTRETIEN DE
L’UNITÉ.
WARNING: THE OXYGEN CONCENTRATORSHOULD NOT BE USED ADJACENT TO OR
STACKED WITH OTHER EQUIPMENT. IF ADJACENT OR STACKED USE IS UNAVOIDABLE,
THE DEVICE SHOULD BE OBSERVED TO VERIFY NORMAL
OPERATION.
WARNING: NO MODIFICATION OF THIS EQUIPMENT ISPERMITTED.
WARNING: USE OF CABLES AND ADAPTERS OTHER THAN
THOSE SPECIFIED, WITH THE EXCEPTION OF CABLES
AND ADAPTERS SOLD BY THE MANUFACTURER OF THE
MEDICAL ELECTRICAL EQUIPMENT AS REPLACEMENTPARTS FOR INTERNAL COMPONENTS, MAY RESULT IN
INCREASED EMISSIONS OF DECREASED IMMUNITY OF
THE OXYGEN CONCENTRATOR.
WARNING: USE ONLY ELECTRICAL VOLTAGE AS SPECIFIEDON THE SPECIFICATION LABEL AFFIXED TO THE DEVICE.
WARNING: DO NOT USE EXTENSION CORDS WITH THISUNIT OR CONNECT TOO MANY PLUGS INTO THE SAMEELECTRICAL OUTLET. THE USE OF EXTENSION CORDS
COULD ADVERSELY AFFECT THE PERFORMANCE OF
THE DEVICE. TOO MANY PLUGS INTO ONE OUTLET CANRESULT IN AN OVERLOAD TO THE ELECTRICAL PANELCAUSING THE BREAKER/FUSE TO ACTIVATE OR FIRE IF THE
BREAKER OR FUSE FAILS TO OPERATE.

CAUTION: Do not position the unit so that itis difficult to access the power cord.
CAUTION: The concentrator should be locatedas to avoid smoke, pollutants or fumes.
CAUTION: Ensure concentrator is operated in an uprightposition.
CAUTION: Always place oxygen supply tubing and powercords in a manner that prevents a trip hazard.
CAUTION: Position the unit away from curtains or drapes,
hot air registers or heaters. Be certain to place the unit
on a flat surface and make sure all sides are at least 1
foot (30 cm) away from a wall or other obstruction. Do
not place the unit in a confined area. Choose a dust andsmoke free-location away from direct sunlight. Do not
operate the unit outdoors unless the unit is plugged
into a Ground Fault Circuit Interrupter (GFCI) protected
outlet.
CAUTION: Do not operate this unit in a restricted orconfined space where ventilation can be limited. This cancause the device to overheat and affect performance.
CAUTION: Do not allow either the air intake or the
air outlet vents to be blocked. DO NOT drop or insert
any object into any openings on the device. This cancause the Oxygen Concentrator to overheat and impair
performance.
CAUTION: The Manufacturer recommends an alternate
source of supplemental oxygen in the event of a power
outage, alarm condition, or mechanical failure. Consult
your physician or Equipment Provider for the type ofreserve system required.
CAUTION: It is very important to select only the prescribed level of oxygen. Do not change the flow selection
unless you have been directed to do so by a licensed
clinician.
CAUTION: The Oxygen Concentrator may be used during
sleep under the recommendation of a licensed clinician.
CAUTION: Operating or storing the Oxygen Concentrator
outside of its normal operating temperature range can
impair the performance of the unit. Refer to the specification section of this manual for storage and operating
temperature limits.
CAUTION: In the event of an alarm or you observe the
Oxygen Concentrator is not working properly; consult
the troubleshooting section of this manual. If you cannot
resolve the problem, consult your Equipment Provider.
CAUTION: If the audio alarm is weak or does not soundat all, consult your Equipment Provider immediately.
CAUTION: For units equipped with the oxygen monitor
option - Contact your Equipment Provider immediately if
the amber OXYGEN MONITOR light remains on for morethan 15 minutes and/or the audio alarm activates.
CAUTION: If the humidifier bottle tubing is not properly
connected to the humidifier bottle fitting or to theoxygen outlet, an oxygen leak can occur.

6 — MN165-1 M

Note: Cannula must be non-kinking, which can be used for a
total length of 25 ft. (7.6 m) max.
Note: Ensure the cannula is fully inserted and secure. You
should hear or feel oxygen flow to the prongs of the nasal
cannula. If oxygen does not seem to flow, first verify thatthe flowmeter ball is registering a flow. Then, place the tip ofthe cannula into a glass of water; if bubbles come out of thecannula, oxygen is flowing. If bubbles do not appear, refer tothe troubleshooting section of this manual.
Note: Always follow the cannula manufacturer’s instructions
for proper use. Replace the disposable cannula as recommended by the cannula manufacturer or your Equipment
Provider. Additional supplies are available from yourEquipment Provider.
Note:Theuseofsomeoxygenadministrationaccessoriesnot
specified for use with this oxygen concentrator may impair
its performance. Recommended accessories are referenced
within this manual.

Note: AirSep offers the OxySafe as an optional accessory.This
is intended to be used in conjunction with the Visionaire 2/3
concentrator.Forcustomersinregionsrequiringcompliance
to EN ISO 8359:1996-Ammendment1:2012, this accessory
will meet this need.
The OxySafe is a thermal fuse to stop the flow of gas in theevent that the downstream cannula or oxygen tubing is
ignited and burns to the OxySafe. It is placed in-line with thenasal cannula or oxygen tubing between the patient and theoxygen outlet of the Visionaire 2/3.
For proper use of the OxySafe, always refer to the manufacturer’s instructions (included with each OxySafe kit).
AirSep offers an OxySafe kit that includes OxySafe with 2”oftubing to connect to the oxygen outlet; PN 20629671

Note:ToEquipmentProvider:ThefollowingoxygenadministrationaccessoriesarerecommendedforusewiththeOxygen
Concentrator• Humidifier Bottle: Part No. HU003-1• Nasal Cannula with 7 feet (2.1 m) of tubing: Part No.
CU002-1• OxySafe Kit: Part No. 20629671
Note:TheManufacturerdoesnotrecommendthesterilizationof this equipment.
Note: If the unit has not been used for an extended time
period, it needs to operate for a minimum of several minutes
before power failure alarm can become activated.
Note: For units equipped with the oxygen monitor option When you turn the unit on, it’s normal for the amber OXYGENMONITOR light to turn on and remain on for up to fiveminutes.
Note: The concentrator releases warm air out the bottom
oftheunitwhichcanpermanentlydiscolortemperaturesensitive flooring surfaces such as vinyl.The concentrator should
not be used over flooring that is sensitive to heat staining.
TheManufacturerisnotresponsibleforflooringthatbecomes
discolored.
Note: To prevent a void warranty, follow all manufacturers’
instructions.
Note: Do not attempt any maintenance other than the possible solutions listed within the manual.
Note:PortableandmobileRFcommunicationsequipmentcan
effect medical electrical equipment.
Note: There is never a danger of depleting the oxygen in aroom when you use your Oxygen Concentrator unit.

VisionAire 2 & 3 User Manual — 7

Introduction
This Patient Manual will acquaint you with AirSep’s
VisionAire 2 and VisionAire 3 Oxygen Concentrators.
Make sure you read and understand all of theinformation contained in this guide before operating
your concentrator. Should you have any questions,
your Equipment Provider will be happy to answer
them for you.

Why Your Physician Prescribed Oxygen
Many people today suffer from a variety of heart,
lung, and other respiratory diseases. A significant
number of these patients can benefit fromsupplemental oxygen therapy for respiratory care athome, in the hospital, or at a medical facility.
Oxygen is a gas that makes up 21% of the room air
we breathe. Our bodies depend on a steady supply
to function properly. Your physician has prescribed
supplemental oxygen therapy because your body is
not able to get enough oxygen from room air. Oxygenis a non-addictive drug, and your physician prescribeda flow sufficient to improve your condition.
Please keep in mind that unauthorized oxygen therapy
can be dangerous. You must seek medical advice
before you use this unit. The Equipment Provider who
supplies your oxygen equipment will demonstrate how
to set the prescribed flow rate.

What is an Oxygen Concentrator?
Oxygen concentrators were introduced in the mid1970s and have become the most convenient, reliable
source of supplemental oxygen available today.
Without an oxygen concentrator, the average patient
would require a delivery of 12 bottles/cylindersof oxygen each month. Your oxygen concentrator
produces all the oxygen you need, with no deliveriesrequired.
The air we breathe contains approximately 21%
oxygen, 78% nitrogen, and 1% other gases. In the
VisionAire 2 or VisionAire 3 unit, room air passes
through a regenerative adsorbent material calledmolecular sieve. This material separates the oxygenfrom the nitrogen and other gases. The result is aconstant supply of high concentration supplemental
oxygen that is delivered to the patient.
Note: There is never a danger of depleting theoxygen in a room when you use your VisionAire 2 or
VisionAire 3 unit.
8 — MN165-1 M

Operator Profile:
AirSep’s Stationary Concentrators are intended
to supply supplemental Oxygen to users sufferingfrom discomfort due to ailments which affect theefficiency of ones lungs to transfer the oxygen in
air to their bloodstream. Users can benefit fromsupplemental oxygen therapy for respiratory care athome, in the hospital, or at a medical facility. Oxygen
Concentrator use requires a physician’s prescription,
and are not intended for life support use.
Although Oxygen therapy can be prescribed forpatients of all ages the typical oxygen therapy patientis older than 65 years of age and suffers from chronic
obstructive pulmonary disease (COPD). Patientstypically have good cognitive abilities and must beable to communicate discomfort. If the user is unable
to communicate discomfort, or unable to read andunderstand the concentrator labeling and instructions
for use, then use is recommended only under thesupervision of one who can. If any discomfort is felt
while using the concentrator, patients are advised
to contact their healthcare provider. Patients arealso advised to have back-up oxygen available (i.e.
cylinder oxygen) in the event of a power outage orconcentrator failure. There are no other unique skills
or user abilities required for concentrator use.

Concentrator Components
Review the figures below and become familiar with
the important parts of your VisionAire 2 or VisionAire
3 Oxygen Concentrator.

Front of Concentrator
Top Handle

Hour Meter

Oxygen Outlet
Side Handle(s)Circuit Breaker Reset Button
Flowmeter
Adjustment
Knob

Low Oxygen
Concentration
Indicator(optional – mandated
for EU and specificcountries)

ON/OFF (I/0) Power Switch

Operating Instructions LabelSpecification Label

Caster(s)Top and Side Handles: Enables convenience incarrying the unit.
On/Off (I/0) Power Switch: Starts and stops theoperation of the unit.

Low Oxygen Concentration Indicator (optional –mandated for EU and specific countries): Whenequipped with an Oxygen Monitor – a warning LED(amber) is activated on the front panel (see ‘Safety
Features’ for details).

Oxygen Outlet: Provides connection for a humidifier(if required) or cannula.

Hour Meter: Records the concentrator’s cumulative
hours of operation.

Circuit Breaker Reset Button: Resets the unit after
electrical overload shutdown.

Operating Instructions Label: Directs usage of theunit.

Flowmeter Adjustment Knob: Controls the oxygen
flow rate in liters per minute (LPM).

Casters: Four casters enable unit to be easily moved,
as needed.
Specification Label: Displays electrical specifications
and serial number.
VisionAire 2 & 3 User Manual — 9

Humidifier Bottle (optional)
Humidifier Bottle
RecessHumidifier Bottle Fitting

Humidifier Bottle Tubing

Back of Concentrator

Humidifier Bottle
Oxygen Outlet
Humidifier Bottle

Oxygen Tubing/CannulaPower Cord

Power Cord: Allows connection of the unit to anelectrical outlet.

Humidifier Bottle (optional): Humidifies the oxygen
before it reaches the patient.

Humidifier Bottle Recess: Area to place the optional
humidifier bottle.

Humidifier Bottle Oxygen Outlet: Connection foroxygen tubing/cannula.

10 — MN165-1 M

Accessories

Safety Features

Recommended for use with VisionAire 2 andVisionAire 3.

Part

Safety Feature

Compressor
Motor

Thermal safety is ensured by athermostat situated in the stator
winding (135°C/ 275 °F). A
safety relief valve is fitted to thecompressor outlet and is calibrated
to 280 kPa (40 psig).

Power
FailureIn the event the unit is operating anda loss of power occurs, an audible
single - pulse intermittent alarm will
activate.

OxygenMonitorThe oxygen monitor detects any dropin concentration below 82% (±3%)
and activates a visual alarm (amber
LED indicator on front panel). If theconcentration continues to operate
below 82%, an audible 2-pulseintermittent alarm will also activate.

Hi Pressure

An over-pressure safety feature
prevents potential damage to device
components at a pressure of 207
kPa (30 psig (± 2 psig). An audible
4-pulse intermittent alarm will
activate.

Low
Pressure

An audible 3-pulse intermittent
alarm will activate if the devicepressure reduces to ≤ 35 kPa (5 psig(± 1 psig).

No Flow
Warning

An obstruction to the flow ofoxygen, such as a pinch or kink in
the delivery cannula, will activate an
audible 6- pulse intermittent alarm.

Product
Filter≥ 10 micron

The use of certain administration accessories which
are not specified for use with this concentrator may
reduce performance and void the manufacturer’s
warranty.
Item

PN

Humidifier Bottle

HU003-1

Cannula, 25 feet (7.6 m)

CU002-4

Cannula, 7 feet (2.1 m)

CU002-1

Humidifier Bottle Tubing

TU176-160

Humidifier Bottle Fitting

F0655-1

Oxygen Tubing,
25 feet (7.6 m)

CU004-3Tubing/Cannula Connector

CU009-1

OxySafe Kit

20629671

Materials in direct or indirect contact
with patient
Item
Concentrator
Cabinet (allparts)
Gas Outlet,
NozzleOn / Off Switch
Hour Meter
Flowmeter
Gas Outlet
Circuit Breaker
Unit Labels
Power Cord
Cord Strain
ReliefCord Wrap
Dust Filter
Humidifier Strap
Humidifier
BottleCannula(Tubing)

Material
NOVA PS6201 Super HighImpact Polystyrene
Aluminum, Black Anodized
DuPont FR50
Acrylonitrile Butadiene Styrene(ABS) Plastic & Acrylic
ABS (STAT-TECH AS1000)
Brass
Button - Melamine or Urea
Formaldehyde, Mounting Bushing– Polyamide 66 (Nylon)
Lexan
Polyvinyl Chloride (PVC), Metal
Nylon
VelcroPolyesterYarn, Rubber
Top – Acrylonitrile Butadiene
Styrene (ABS)
Bottle – Polypropylene (PP)
Diffuser -Polyvinyl Chloride(PVC)
Polyvinyl Chloride (PVC)

VisionAire 2 & 3 User Manual — 11

Operating Instructions
Review the following information before you operate
your oxygen concentrator.
Note: The concentrator releases warm air out the bottom
oftheunitwhichcanpermanentlydiscolortemperaturesensitive flooring surfaces such as vinyl.The concentrator should
not be used over flooring that is sensitive to heat staining.
Themanufacturerisnotresponsibleforflooringthatbecomes
discolored.

Humidifier Bottle Connection (Optional)
If additional humidification is required with youroxygen therapy, perform the following steps each time
you fill or clean the humidifier, which may have been
initially set up for your use.
1. Remove the humidifier bottle from the humidifier
bottle recess.
2. Open the humidifier bottle. If you have a pre-filled
bottle, do not perform this step. Proceed to step 5.
3. Fill the humidifier bottle with cool or cold water(distilled water is preferred) to the fill line indicatedon the bottle. DO NOT OVERFILL.
4. Re-connect the top cover to the humidifier bottle.
5. Place the humidifier bottle in the humidifier bottle
recess on the back of the concentrator and connect
the humidifier bottle tubing to the oxygen outlet
and the humidifier bottle fitting.
CAUTION: If the humidifier bottle tubing is
not properly connected to the humidifier
bottle fitting or to the oxygen outlet, an
oxygen leak can occur.

Cannula Connection
Connect the tubing and cannula to the unit’s oxygen
outlet, or to the optional humidifier’s oxygen outlet.

Starting the Concentrator
WARNING: “NO SMOKING – OXYGEN IN USE”
SIGNS MUST BE PROMINENTLY DISPLAYEDIN THE HOME, OR WHERE OXYGEN IS IN USE.
PATIENTS AND THEIR CAREGIVERS MUST BE
INFORMED ABOUT THE DANGERS OF SMOKING IN THE
PRESENCE OF, OR WHILE USING, MEDICAL OXYGEN.
WARNING: DO NOT USE EXTENSION CORDS WITH THISUNIT OR CONNECT TOO MANY PLUGS INTO THE SAMEELECTRICAL OUTLET. THE USE OF EXTENSION CORDS
COULD ADVERSELY AFFECT THE PERFORMANCE OF
THE DEVICE. TOO MANY PLUGS INTO ONE OUTLET CANRESULT IN AN OVERLOAD TO THE ELECTRICAL PANELCAUSING THE BREAKER/FUSE TO ACTIVATE OR FIRE IF THE
BREAKER OR FUSE FAILS TO OPERATE.
CAUTION: Position the unit away from curtains
or drapes, hot air registers or heaters. Be certain
to place the unit on a flat surface and make sure
all sides are at least 1 foot (30 cm) away from a wall orother obstruction. Do not place the unit in a confinedarea. Choose a dust and smoke free-location away fromdirect sunlight. Do not operate the unit outdoors unlessthe unit is plugged into a Ground Fault Interrupter (GFI)
protected outlet.

1. Locate the unit near the electrical outlet in theroom where you spend most of your time.
2. Insert the power cord plug into the electrical outlet.
3. Set the I/0 power switch to the “I” position to turnthe unit on. An audible alarm will loudly sound forapproximately 5 seconds.
4. The low oxygen concentration indicator remains
on for a few minutes and until the oxygen
concentration reaches minimum concentration
requirements, (only pertains to unit equipped withan Oxygen Monitor)
5. To set the flow of supplemental oxygen, turn theflowmeter adjustment knob left or right until theball inside the flowmeter centers on the flow line
number prescribed by your physician. To viewthe flowmeter at the proper angle, note that theback line and the front number line must give
appearance of one line.
6. The concentrator is now ready for use.
Note: Optimal oxygen concentration is obtained within 10
minutes after the device is switched on (90% of the concentration is obtained after approximately 5 minutes).

12 — MN165-1 M

CAUTION: It is very important to select only theprescribed level of oxygen. Do not change theflow selection unless you have been directed to
do so by a licensed clinician.

Normally, you should not need to adjust the flowmeteron your unit. If you turn the flowmeter adjustment
knob clockwise, you will decrease and can shut off theflow of oxygen from your unit.
Note: Ensure the cannula is fully inserted and secure. You
should hear or feel oxygen flow to the prongs of the nasal
cannula. If oxygen does not seem to flow, first verify thatthe flowmeter ball is registering a flow. Then, place the tip ofthe cannula into a glass of water; if bubbles come out of thecannula, oxygen is flowing. If bubbles do not appear, refer tothe troubleshooting section of this manual.
CAUTION: Ensure concentrator is operated in
an upright position.

Turning the Concentrator Off
Set the I/0 power switch to the “0” position to turn off
the unit.

Care and Cleaning

WARNING: ELECTRICAL SHOCK HAZARD.
TURN OFF THE UNIT AND DISCONNECT THE
POWER CORD FROM THE ELECTRIC OUTLETBEFORE YOU CLEAN THE UNIT TO PREVENT
ACCIDENTAL ELECTRICAL SHOCK AND BURN HAZARD.
ONLY YOUR EQUIPMENT PROVIDER OR A QUALIFIEDSERVICE TECHNICIAN SHOULD REMOVE THE COVERS ORSERVICE THE UNIT.
WARNING: CARE SHOULD BE TAKEN TO PREVENT THE
OXYGEN CONCENTRATOR FROM GETTING WET OR
ALLOWING FLUIDS TO ENTER THE UNIT. THIS CAN CAUSETHE UNIT TO MALFUNCTION OR SHUT DOWN, AND CAUSEAN INCREASED RISK FOR ELECTRICAL SHOCK OR BURNS.
WARNING: DO NOT USE LIQUID DIRECTLY ON THE UNIT.
A LIST OF UNDESIRABLE CHEMICAL AGENTS INCLUDES
BUT IS NOT LIMITED TO THE FOLLOWING: ALCOHOL
AND ALCOHOL-BASED PRODUCTS, CONCENTRATEDCHLORINE-BASEDPRODUCTS(ETHYLENECHLORIDE),AND
OIL-BASED PRODUCTS (PINE-SOL®, LESTOIL®). THESE
ARE NOT TO BE USED TO CLEAN THE PLASTIC HOUSING
ON THE UNIT, AS THEY CAN DAMAGE THE UNIT’S PLASTIC.
WARNING: CLEAN THE CABINET, CONTROL PANEL, AND
POWER CORD ONLY WITH A MILD HOUSEHOLD CLEANERAPPLIED WITH A DAMP CLOTH OR SPONGE, AND THEN
WIPE ALL SURFACES DRY. DO NOT ALLOW ANY LIQUID TOGET INSIDE THE DEVICE.
Note: Always follow the cannula manufacturer’s instructions
for proper use. Replace the disposable cannula as recommended by the cannula manufacturer or your Equipment
Provider. Additional supplies are available from yourEquipment Provider.
Note: To prevent a voided AirSep warranty, follow all manufacturers’ instructions.
Note: AirSep does not recommend the sterilization of thisequipment.

Humidifier Bottle (optional)
Check water level daily and add water as needed
To clean and disinfect the humidifier, follow yourEquipment Provider’s instructions, or the instructions
included with the humidifier bottle.

Outside Cabinet
Use a mild household cleaner applied with adamp cloth or sponge to clean the exterior of theconcentrator, and then wipe all surfaces dry. Do notallow any liquids to enter the concentrator.

VisionAire 2 & 3 User Manual — 13

Troubleshooting
If your concentrator fails to operate properly, consult your Equipment Provider, and refer to the troubleshooting
chart on the following pages for probable causes and solutions.
Note: Do not attempt any maintenance other than the possible solutions listed within the manual
Note: If the unit has not been used for an extended time period, it needs to operate for several minutes before power failure alarm
can become activated.

Problem

Probable Cause

Solution

Unit does
not operate.
Power failure
condition
causes an
alarm tosound.

The power cord is notconnected to the electricaloutlet.

Check power cord at the electrical outlet for a proper
connection.

No power at the electricaloutlet.

Check power source, wall switch, in-house fuse or circuit
breaker.

Circuit breaker on
concentrator unit isactivated.

Press (do not hold in) the circuit breaker reset button in thefront of the unit.

Dirty or obstructed
humidifier bottle, or leakpresent.

Remove the humidifier bottle, and if flow is restored, clean
or replace the humidifier bottle.

Defective nasal cannula, orother oxygen accessories,
i.e. oxygen tubing.

Remove and check accessories for kinks or obstructions.
Replace if needed.

Limited or nooxygen flow.

If the circuit breaker trips again or the alarm continues tosound after the unit is turned on, contact your Equipment
Provider.

Cannula tubing loose.

Check cannula tubing connection at control panel.

Condensationcollects in theoxygen tubing
when you use
a humidifier
bottleUnit not properly
ventilated.
Elevated operating
temperatures.

Make sure unit is positioned away from curtains or drapes,
hot air registers or heaters. Be certain to place the unit so allsides are at least 1 foot (30 cm) away from a wall or otherobstruction. Do not place the unit in a confined area.
Refill humidifier bottle with COLD water.
DO NOT OVERFILL. Allow oxygen tubing to dry out, or
replace with new tubing.

Intermittent
alarm sounds
at one second
intervals.

See “Safety Features” fora description of auditory
indicators.

Set I/0 power switch to 0 position, use your reserve oxygen
supply (if provided), and consult your Equipment Provider
immediately.

OxygenConcentration
Indicator
remains lit(amber)Oxygen concentration is ≤82% (±3%).

Set I/0 power switch to the 0 position, use your reserve
oxygen supply (if provided), and consult your Equipment
Provider immediately.

All other
problems.

14 — MN165-1 M

Set I/0 power switch to the 0 position, use your reserve
oxygen supply (if provided), and consult your Equipment
Provider immediately.

Product Specifications
For correct use of the device refer to the following
chart. The chart provides important information
about the recommended operating environments, or
operating conditions.

Specifications
Flow
Specifications(1)1/2 lpm – 3 lpm
1/8 lpm – 2 lpm

OxygenConcentration(2)

90% (+ 5.5% / - 3%)

Electrical
Requirements

115 VAC / 60 Hz, 3.0 A
230 VAC / 50 Hz, 1.5 A
230 VAC/ 60 Hz, 1.0 A

PowerConsumption
Sound
Dimensions

VisionAire 2 = 200 Watts (+/- 10%)
VisionAire 3 = 200 Watts (+/- 10%)45 dba anechoic
14.1 in. W x 11.5 in. D x 20.8 in. H(35.8 cm W x 29.2 cm D x 52.8 cm H)

Weight30 lbs (13.6 kg)

Electrical Shock
Protection

Class II, Type BF

Environmental
Limit
Conditions(3)Operating conditions: 5°C to 40°C (41°F
to 104°F) at altitudes up to 10,000 ft.
(523 mmHg) above sea level.
Storage: -20°C to 60°C (-4°F to 140°F)
Relative Humidity: up to 95% RH
non-condensing(1) ±10% of indicated setting, or 200 mL, whichever is greater(2) Performance based at 21°C (70°F) at nominal operatingpressure range: 0-7 kPa (0-1 psig back pressure). Maximum
outlet pressure : 55 kPa. ( 7.98 psig)(3) Operating outside of these operational specifications canlimit the concentrators ability to meet Oxygen Concentration
specification at higher liter flow rates.

VisionAire 2 & 3 User Manual — 15

Medical equipment needs special precautions regarding EMC and need to be installed and put into service
according to the EMC information provided in this section.
Guidance and Manufacturer’s Declaration—Electromagnetic Immunity
The unit is intended for use in the electromagnetic environment specified below.The customer or the user of theunit should assure that it is used in such an environment.
Immunity test

IEC 60601 test level

Compliance
level

Electromagnetic
environment –guidance±6 kV contact±6 kV contact±8 kV air

±8 kV air

±2 kV for power supply
lines±2 kV forpower supplylines

IEC 61000-4-2
Electrical fasttransient/burst
IEC 61000-4-4

Surge
IEC 61000-4-5±1 kV for input/output
lines±1 kV line(s) to line(s)Electromagnetic environment – guidance
Floors should be wood, concrete or ceramic tile. If floors are covered with synthetic
material, the relative humidity should be atleast 30 %.

Mains power quality should be that of a typicalcommercial or hospital environment.

N/A±1 kV line(s)to line(s)±2 kV line(s) to earth

Mains power quality should be that of a typicalcommercial or hospital environment.

<5 % UT<5 % UT(>95 % dip in UT )for 0,5 cycle
40 % UT
Voltage dips, shortinterruptions andvoltage variations
on power supply
input lines IEC
61000-4-11(>95 % dip
in UT )
for 0,5 cycle
40 % UT(60 % dip in UT )(60 % dip in
UT )for 5 cycles

for 5 cycles

70 % UT

70 % UT(30 % dip in UT )(30 % dip in
UT )for 25 cycles<5 % UT(>95 % dip in UT )for 5 sec

Mains power quality should be that of a typicalcommercialorhospitalenvironment.Iftheuser
oftheunitrequirescontinuedoperationduring
powermainsinterruptions,itisrecommended
that the unit be powered from an uninterruptible power supply or a battery.

for 25 cycles<5 % UT(>95 % dip
in UT )
for 5 sec

Power frequency (50/60 Hz)
magnetic field IEC
61000-4-8

3A/m

3A/m

Power frequency magnetic fields should be atlevels characteristic of a typical location in a
typical commercial or hospital environment.

NOTE UT is the A.C. mains voltage prior to application of the test level.

16 — MN165-1 M

Guidance and Manufacturer’s Declaration–Electromagnetic Immunity
The unit is intended for use in the electromagnetic environment specified below.The customer or the user of theunit should assure that it is used in such an environment.
Immunity test

IEC 60601 testlevel

Compliance
level

Electromagnetic environment – guidance

Conducted RF

3 Vrms3 VrmsIEC 61000-4-6

150 kHz to 80
MHz

PortableandmobileRFcommunicationsequipment
should be used no closer to any part of theVisionAire,includingcables,thantherecommendedseparation
distance calculated from the equation applicable tothe frequency of the transmitter.

Radiated RF
IEC 61000-4-3

3 V/m
80 MHz to 2,5
GHz

3 V/m
Recommended separation distanced  1.2 P
d  1.2 P = 80 MHz to 800 MHz
d  1.2 P = 800 MHz to 2,5 GHz
Where P is the maximum output power rating of thetransmitter in watts (W) according to the transmitter
manufacturer and d is the recommended separation
distance in metres (m).
Field strengths from fixed RF transmitters, as determined by an electromagnetic site survey,a shouldbe less than the compliance level in each frequency
range. b
Interference may occur in the vicinity of equipment
marked with the following symbol:

NOTE 1 At 80 MHz and 800 MHz, the higher frequency range applies.
NOTE2Theseguidelinesmaynotapplyinallsituations.Electromagneticpropagationisaffectedbyabsorptionandreflection from structures, objects and people.
aField strengths from fixed transmitters, such as base stations for radio (cellular/cordless) telephones and land
mobileradios,amateurradio,AMandFMradiobroadcastandTVbroadcastcannotbepredictedtheoreticallywith
accuracy.To assess the electromagnetic environment due to fixed RF transmitters, an electromagnetic site survey
should be considered. If the measured field strength in the location in which the VisionAire is used exceeds theapplicable RF compliance level above, theVisionAire should be observed to verify normal operation. If abnormal
performanceisobserved,additionalmeasuresmaybenecessary,suchasre-orientingorrelocatingtheVisionAire.
b

Over the frequency range 150 kHz to 80 MHz, field strengths should be less than 3 V/m.

VisionAire 2 & 3 User Manual — 17

Recommended separation distances
between portable and mobile RF communications equipment and the unit
The unit is intended for use in an electromagnetic environment in which radiated RF disturbances are controlled.
The customer or the user of the unit can help prevent electromagnetic interference by maintaining a minimum
distancebetweenportableandmobileRFcommunicationsequipment(transmitters)andtheunitasrecommended
below, according to the maximum output power of the communications equipment.
Rated maximum
output power oftransmitter

Separation distance according to frequency of transmitterm
150 kHz to 80 MHz

80 MHz to 800 MHz

800 MHz to 2,5 GHzW

d  1.2 P

d  1.2 P

d  1.2 P

0,01

0.12

0.12

0.23

0,1

0.38

0.38

0.73

1

1.2

1.2

2.3

10

3.8

3.8

7.3

100

12

12

23

For transmitters rated at a maximum output power not listed above, the recommended separation distance d inmetres (m) can be estimated using the equation applicable to the frequency of the transmitter, where P is the maximum output power rating of the transmitter in watts (W) according to the transmitter manufacturer.
NOTE 1 At 80 MHz and 800 MHz, the separation distance for the higher frequency range applies.
NOTE2Theseguidelinesmaynotapplyinallsituations.Electromagneticpropagationisaffectedbyabsorptionandreflection from structures, objects and people.

Guidance and Manufacturer’s Declaration­—Electromagnetic Emissions

The unit is intended for use in the electromagnetic environment specified below.The customer or the user of theunit should assure that it is used in such an environment.
Emissions Test

RF emissions
CISPR 11

RF emissions
CISPR 11

ComplianceElectromagnetic Environment - GuidanceGroup 1The unit uses RF energy only for its internal function. Therefore, its RF
emissions are very low and are not likely to cause any interference innearby electronic equipment.

Class B

Harmonicemissions
IEC 61000-3-2Class A

Voltagefluctuations/
flicker emissions
IEC 61000-3-3Complies

18 — MN165-1 M

The unit is suitable for use in all establishments, including domestic establishmentsandthosedirectlyconnectedtothepubliclow-voltagepower supply network that supplies buildings used for domestic purposes.

Conformity with EN 60-601 (§ 6.8.2 b) /Classification:

Disposal“The manufacturer, assembler, installer or importerare not considered to be responsible themselves
for the consequences on the safety, reliability andcharacteristics of a device unless:- The assembly, fitting, extensions, adjustments,
modifications or repairs have been performed bypersons authorized by the party in question,- The electrical installation of the corresponding
premises complies with IEC regulations.
- The device is used in accordance with the
instructions for use. “If the replacement parts used
for the periodic servicing by an approved technician
do not comply with the manufacturer’s specifications,the latter is absolved from all responsibility in theevent of an accident. Do not open the device while
in operation: risk of electrical shock. This device
complies with the requirements of the MDD (93/42/
EEC:2007/47/EC) Annex I European directive but
its operation may be affected by other devices being
used close by, such as diathermy and high frequency
electro-surgical equipment, defibrillators, short wave
therapy equipment, mobile telephones, CB and otherportable devices, microwave ovens, induction plates
or even remote control toys and more generallyelectromagnetic interferences which exceed the levels
specified by the EN 60601-1-2 standard.

Method of disposing of waste: All waste from
AirSep’s VisionAire 2 and VisionAire 3 Oxygen
Concentrator must be disposed of using theappropriate methods specified by local authorities.
Method for disposing of the device: In order topreserve the environment, the concentrator must be
disposed of using the appropriate methods specified
by local authorities.
AVERTISSEMENT: CET APPAREIL PRODUIT
DE L’OXYGÈNE À CONCENTRATION ÉLEVÉE,FAVORISANT UNE COMBUSTION RAPIDE.
NE PAS PERMETTRE DE FUMER OU DES
FLAMMES NUES DANS LA MÊME CHAMBRE: (1) CETAPPAREIL OU (2) TOUT ACCESSOIRE CONTENANT DE L’OXYGÈNE. NE PAS UTILISER DE PRODUITS À BASE D’HUILE,DE GRAISSE OU DE PÉTROLE SUR OU À PROXIMITÉ DE
L’UNITÉ. DÉCONNECTER LE CORDON D’ALIMENTATION DE
LA PRISE ÉLECTRIQUE AVANT DE NETTOYER OU DE FAIRE
L’ENTRETIEN DE L’UNITÉ.
AVERTISSEMENT: RISQUE DE CHOC ÉLECTRIQUE. NE
PAS ENLEVER LES COUVERCLES LORSQUE L’UNITÉ ESTBRANCHÉE. SEULS VOTRE FOURNISSEUR D’ÉQUIPEMENTOU UN TECHNICIEN DE SERVICE QUALIFIÉ DEVRAITENLEVER LES COUVERCLES OU FAIRE L’ENTRETIEN DE
L’UNITÉ.

Classification
Type of protection against electric shock:Class II Protection from electric shock is achieved by
DOUBLE INSULATION.
Protective earthing or reliance upon installation
conditions are not required.
Degree of protection against electric shock:Type BF Equipment providing a particular degree ofprotection against electric shock, particularly
regarding:1) Allowable leakage current;
2) Reliability of protective earth connection(if present).
Not intended for direct cardiac application.
Method of cleaning and infection control allowed:
Please refer to your Equipment Provider and the
VisionAire Service Manual.
Degree of safety of application in the presence offlammable gases:
Equipment not suitable for such application.
Mode of operation:
Continuous duty.
VisionAire 2 & 3 User Manual — 19

CAIRE Inc.
2200 Airport Industrial Dr., Ste. 500
Ball Ground, GA 30107
1-800-482-2473
Worldwide: www.chartindustries.com/Respiratory-Healthcare

Medical Product Services GmbH
Borngasse 20
35619 Braunfels, Germany
info@mps-gmbh.eu
Worldwide: www.chartindustries.com/Respiratory-Healthcare

As of the date of this publication, VisionAire, AirSep and their related logos
Copyright © 2016 Chart Industries. CAIRE Inc. reserves
the right to discontinue its products, or change the prices,materials,equipment,quality,descriptions,specifications
and/or processes to its products at any time without priornotice and with no further obligation or consequence.
All rights not expressly stated herein are reserved by us,
as applicable.

(as applicable) are trademarks registered by CAIRE Inc.
Lysol® is a registered trademark of Reckitt Benckiser, UK.
OxySafe® is a registered trademark of Applied Home Healthcare Equipment.
Pine-Sol® and Lestoil® are registered trademarks of The Clorox Company.

|MN165-1_MN~

MN165-1_M