PT0-13521

ポータブルマルチガスモニター

ＧＸ－６０００
取扱説明書（PT0-135）〒174-8744 東京都板橋区小豆沢 2-7-6
ホームページ https://www.rikenkeiki.co.jp/

目次
1 製品のアウトライン ........................................................................................................................................... 3
はじめに ........................................................................................................................................................... 3
使用目的 .......................................................................................................................................................... 3
検知対象ガスの確認方法 ................................................................................................................................. 4
危険、警告、注意、注記の定義 ......................................................................................................................... 5
規格および防爆仕様の確認方法 ...................................................................................................................... 5
2 安全上、大切なお知らせ................................................................................................................................... 6
2-1. 危険事項.................................................................................................................................................. 6
2-2. 警告事項.................................................................................................................................................. 8
2-3. 注意事項.................................................................................................................................................. 9
2-4. セーフティーインフォメーション................................................................................................................. 12
3 製品の構成 .................................................................................................................................................... 14
3-1. 本体および標準付属品........................................................................................................................... 14
3-2. 各部の名称と働き .................................................................................................................................. 18
4 警報動作........................................................................................................................................................ 224-1. ガス警報動作 ......................................................................................................................................... 22
4-2. 故障警報動作 ........................................................................................................................................ 25
4-3. パニック警報 .......................................................................................................................................... 26
4-4. マンダウン警報....................................................................................................................................... 27
5 使用方法........................................................................................................................................................ 28
5-1. ご使用するにあたって ............................................................................................................................. 28
5-2. 始動準備................................................................................................................................................ 28
5-3. 始動方法................................................................................................................................................ 36
5-4. ゼロ校正 ................................................................................................................................................ 39
5-5. プレフィルターチューブの校正コード入力 ................................................................................................ 44（VOC<10.0eV>センサを搭載している仕様のみ） ........................................................................................... 44
5-6. 検知する ................................................................................................................................................ 46
5-7. 電源を切る ............................................................................................................................................. 53
6 設定方法........................................................................................................................................................ 54
6-1. DISP モードのフロー ............................................................................................................................... 54
6-2. 各種設定................................................................................................................................................ 59
6-3. ユーザーモード設定 ............................................................................................................................... 75
7 保守点検........................................................................................................................................................ 82
7-1. 点検の頻度と点検項目........................................................................................................................... 82
7-2. 清掃方法................................................................................................................................................ 84
7-3. 各部品の交換 ........................................................................................................................................ 85
8 保管および廃棄について ................................................................................................................................ 87
8-1. 保管または長期使用しない場合の処置 .................................................................................................. 87
8-2. 再度使用する場合の処置 ....................................................................................................................... 87
8-3. 製品の廃棄 ............................................................................................................................................ 88
9 トラブルシューティング .................................................................................................................................... 89
9-1. 機器の異常 ............................................................................................................................................ 89
9-2. 指示値の異常 ........................................................................................................................................ 91
10 製品仕様 ..................................................................................................................................................... 92
10-1. 仕様一覧.............................................................................................................................................. 92
10-2. 付属品一覧 .......................................................................................................................................... 96
11 付録 ............................................................................................................................................................. 97
11-1. 校正履歴機能／各種トレンド機能／イベント履歴機能 ........................................................................... 97
11-2. 用語の定義 .......................................................................................................................................... 99
11-3. VOC 読み替えガスリスト .................................................................................................................... 100-2-1 製品のアウトライン

はじめに

1
1

製品のアウトライン

はじめに
この度は、ポータブルマルチガスモニターGX-6000（以降「本器」）をお買い上げいただき、誠にありがとうございます。まず、お買い求めの製品型番と本取扱説明書が対象とする製品型番が一致することをご確認ください。
この取扱説明書は、本器を正しくご使用いただくための取扱方法と仕様が記載されています。本器を初めてご使用になる方、および既にご使用経験のある方も本書をよくお読みいただき、内容を理解した上で実際にご使用ください。
なお、製品改良のために、この説明書の内容を将来予告なしに変更することがあります。また、この説明書の全部または一部を無断で複写または転載することを禁じます。
保証期間の内外を問わず本器をご使用することによって生じたいかなる事故および損害の補償はいたしま
せん。
保証書に記載される保証規定を必ずご確認ください。

使用目的
本器は、最大 6 種類のセンサで 6 種類のガスを 1 台で検知できるポンプ吸引式のマルチガスモニターです。
搭載できるセンサは、ベースとなる酸素、可燃性ガス＜%LEL＞、毒性ガス（一酸化炭素・硫化水素）センサの他、揮発性有機化合物や二酸化硫黄などをはじめとする複数種類のスマートセンサから最大 2 種類お選び
いただけます。
本器が検知する可燃性ガスは、一般的な工場やオイルタンカーなどで使用する一般可燃性ガス HC（イソブタン換算表示）または CH4（メタン）です。
本器は、検知結果により、生命・安全の保障をするものではありません。
本器は搭載するセンサにより検知対象ガスが異なります。ご使用前に検知対象ガスを再度ご確認いただき、目的に応じた正しいガス検知を行ってください。
お手元の GX-6000 の検知対象ガスについては、「検知対象ガスの確認方法」（P.4）より、確認してください。

-3-

1 製品のアウトライン

検知対象ガスの確認方法

検知対象ガスの確認方法
本器は、搭載するセンサにより検知対象ガスの組み合わせが異なります。
ご使用前に、製品の側面に貼付された銘板から、お手元の GX-6000 の検知対象ガスを確認してください。

M1

製品コードから検知対象ガスを確認
Check the gases to be detected with the product code
Fixed

Fixed

⑨
＜ベースとなるセンサ＞
記載位置

仕様

記号
H：あり（HC）①

可燃性ガス（HC/CH4）＜%LEL＞センサ

M：あり（CH4）
0：なし②

酸素（O2）センサ③

硫化水素（H2S）センサ④

一酸化炭素（CO）センサ

1：あり
0：なし
1：あり
0：なし
1：あり
0：なし

＜スマートセンサ＞
記載位置

⑤⑥
⑦⑧

仕様

記号

揮発性有機化合物（VOC）＜10.6eV / ppb＞センサ

P1

揮発性有機化合物（VOC）＜10.6eV / ppm＞センサ

P2

揮発性有機化合物（VOC）＜10.0eV＞センサ

P3

二酸化硫黄（SO2）センサ

E1

二酸化窒素（NO2）センサ

E2

シアン化水素（HCN）センサ

E3

アンモニア（NH3）センサ

E4

塩素（Cl2）センサ

E5

ホスフィン（PH3）センサ

E6

二酸化炭素（CO2）＜vol%＞センサ

D1

可燃性ガス（HC）＜%LEL / vol%＞センサ

D2

可燃性ガス（CH4）＜%LEL / vol%＞センサ

D3

二酸化炭素（CO2）＜ppm＞センサ

D4
00

―

-4-

1 製品のアウトライン

危険、警告、注意、注記の定義＜バッテリ＞
記載位置

仕様

記号
L

リチウムイオン電池⑨

D
アルカリ乾電池例） 検知対象ガス「M110P1E3L」と表記されている場合は、検知対象ガスは「CH4＜%LEL＞、O2、
H2S、VOC（ppb）、HCN」、またバッテリ仕様は「リチウムイオン電池」となります。

危険、警告、注意、注記の定義
本取扱説明書では、安全かつ効果的な作業が行えるように、次の見出しを使用しています。

危険取り扱いを誤った場合、「人命、人体または物に重大な被害を及ぼすことが想定される」ということを意味します。

警告

取り扱いを誤った場合、「身体または物に重大な被害を及ぼすことが想定される」ということを意味します。

注意

取り扱いを誤った場合、「身体または物に軽微な被害を及ぼすことが想定される」ということを意味します。

注記

取り扱い上のアドバイスを意味します。

規格および防爆仕様の確認方法
本器は、規格や防爆検定の種類によって仕様が異なります。ご使用になる前に、お手元にある製品の仕様をご確
認 く だ さ い 。 な お 、 CE/UKCA マ ー キ ン グ 仕 様 を ご 使 用 に な る 場 合 、 巻 末 の 自 己 宣 言 書 （ Declaration ofConformity）を参照してください。
製品の仕様は、製品に下図の通り貼付された銘板よりご確認いただけます。
上側

右側面

ATEX/IECEx/UKEX 検定銘板
CE/UKCA マーク

Japan Ex 検定銘板

銘板貼付位置-5-

2 安全上、大切なお知らせ

2-1. 危険事項

2
2

安全上、大切なお知らせ

本器の性能を維持し、安全にお使いいただくため、以下の危険、警告、注意事項を守ってください。

2-1. 危険事項
危険防爆に関してメインユニットの防爆に関して
回路・構造などの改造または変更は、行わないでください。

酸素濃度の測定では、空気と可燃性ガスまたは蒸気と毒性ガスの混合物以外の測定をしないでください。
 本器を携帯して危険な場所で使用する場合は、静電気の帯電による危険防止総合対策として、① 使用する衣服は帯電防止作業服、履き物は導電性履き物（帯電防止作業靴）を使用する② 屋内では、導電性作業床（漏洩抵抗 10MΩ以下）の環境で使用する
メインユニット 電池ユニット
としてください。
 メインユニットの合格番号は、搭載するスマートセンサの検知原理によって異なります。お買い求めの製品に搭載されたスマートセンサの検知原理と合格番号の組合せが正しいことをご確認の上、ご使用ください。
搭載するスマートセンサの検知原理と合格番号について搭載するスマートセンサの検知原理
赤外線式のみ赤外線式＋光イオン化（PID）式
赤外線式＋定電位電解式
赤外線式以外、またはスマートセンサなし


合格番号TC21155
TC21154
TC21153

接続できる電池ユニットは、BUL-6000(合格番号 TC21156)または BUD-6000(合格番号 TC21157)です。
本体ユニットの定格は以下の通りです。
スマートセンサ 1 回路 ：許容電圧 4.95V、許容電流 0.770A、許容電力 787mW
スマートセンサ 2 回路 ：許容電圧 4.95V、許容電流 0.770A、許容電力 787mW
ポンプ回路：許容電圧 4.95V、許容電流 0.770A、許容電力 787mW
ブザー回路：許容電圧 4.95V、許容電流 0.355A、許容電力 363mW
モーター回路：許容電圧 4.95V、許容電流 0.209A、許容電力 214mW
メイン回路：許容電圧 4.95V、許容電流 1.112A、許容電力 1137mW
周囲温度：-20℃～+50℃
 メインユニットの防爆等級は ExiaⅡCT4X です。

-6-

2 安全上、大切なお知らせ

2-1. 危険事項

危険防爆に関して電池ユニットの防爆に関して
 回路・構造などの改造または変更は、行わないでください。
本器を携帯して危険な場所で使用する場合は、静電気の帯電による危険防止総合対策として、① 使用する衣服は帯電防止作業服、履き物は導電性履き物（帯電防止作業靴）を使用する。
② 屋内では、導電性作業床（漏洩抵抗 10MΩ以下）の環境で使用するとしてください。

接続できるメインユニットは、GX-6000（合格番号 TC21153、TC21154、TC21155）です。

リチウムイオン電池ユニット BUL-6000 の定格は以下の通りです。
スマートセンサ 1 回路 ：最大電圧 4.2V、最大電流 0.653A、最大電力 561mW
スマートセンサ 2 回路 ：最大電圧 4.2V、最大電流 0.653A、最大電力 561mW
ポンプ回路：最大電圧 4.2V、最大電流 0.653A、最大電力 561mW
ブザー回路：最大電圧 4.2V、最大電流 0.302A、最大電力 259mW
モーター回路：最大電圧 4.2V、最大電流 0.177A、最大電力 152mW
メイン回路：最大電圧 4.2V、最大電流 0.943A、最大電力 811mW
充電端子：許容電圧 17.8V、許容電流 2.72A
周囲温度：-20℃～+50℃

リチウムイオン電池ユニット BUL-6000 の防爆等級は ExiaⅡCT4 です。

乾電池ユニット BUD-6000 の定格は以下の通りです。
スマートセンサ 1 回路 ：最大電圧 4.95V、最大電流 0.770A、最大電力 787mW
スマートセンサ 2 回路 ：最大電圧 4.95V、最大電流 0.770A、最大電力 787mW
ポンプ回路：最大電圧 4.95V、最大電流 0.770A、最大電力 787mW
ブザー回路：最大電圧 4.95V、最大電流 0.355A、最大電力 363mW
モーター回路：最大電圧 4.95V、最大電流 0.209A、最大電力 214mW
メイン回路：最大電圧 4.95V、最大電流 1.112A、最大電力 1137mW
電源：DC4.5V 150mA （LR6 株式会社東芝 3 本）
周囲温度：-20℃～+50℃

乾電池ユニット BUD-6000 の防爆等級は ExiaⅡCT4 です。
ご使用において
マンホールの中や密閉された場所を測定する場合、絶対にマンホールの入り口に身を乗りだしたり、中をのぞき込まないでください。酸素欠乏空気、その他のガスが吹き出す可能性があり危険です。
ガス排出口には酸素欠乏空気などが排出される場合があります。絶対に吸気しないでください。
ガス排出口から高濃度（100%LEL 以上）のガスが排出される場合があります。絶対に火気を近づけないでください。

警告

万一、本器に異常が見つかった場合は、速やかに販売店または最寄りの弊社営業所までご連絡ください。最寄りの営業所につきましては、弊社ホームページよりご確認ください。
ホームページ https://www.rikenkeiki.co.jp/-7-2 安全上、大切なお知らせ

2-2. 警告事項

2-2. 警告事項
警告サンプリングポイントの圧力
本器は大気圧状態の雰囲気のガスを吸引するように作られています。本器のガス吸入口、ガス排出口に過大な圧力をかけると、内部から検知ガスが漏洩する可能性があり危険です。過大な圧力がかからないようにして使用してください。

センサの取り扱い
本器内の定電位電解式センサ、ガルバニ電池式センサは絶対に分解しないでください。内部の電解液が皮膚に触れると、皮膚がただれる恐れがあります。また眼に入ると失明する恐れがあります。衣服に付着した場合には、変色したり、穴が開く恐れがあります。万一、電解液に触れた場合は、触れた部分
を直ちに水で十分洗浄してください。

周辺空気でのエア調整
エア調整を周辺空気で行う場合は、周辺が新鮮な大気であることを確認してから行ってください。雑ガスなどが存在する状態で行うと、正しい調整が行えず、実際にガスが漏洩した場合、正しく検知できず危険です。

ガス警報が出たときの対応
ガス警報を発した場合は大変危険です。お客様の判断により適切な処置を行ってください。
パニック警報、マンダウン警報について
パニック警報およびマンダウン警報は、使用者または周囲の人の判断を補助するものであり、生命・安全の保障をするものではありません。本機能だけに頼った運用は、行わないでください。
（マンダウン警報は、通常 OFF 設定となっており、使用できません。本機能をご利用になる場合は、販売店または最寄りの弊社営業所までご連絡ください。）

パニック警報およびマンダウン警報が発報された場合、周囲の人は周りの状況を確認した上で、適切に処置してください。
電池残量の確認
ご使用前に電池残量を確認してください。初めて使用する前および長期間使用しなかった場合は、電池が消耗していることが考えられます。新しい電池に交換してからご使用ください。

電池電圧低下警報が発せられると、ガス検知を行えなくなります。使用中に発報した場合は、電源を切り、安全な場所で速やかに充電または電池を交換してください。
その他

火中に投げ入れないでください。

洗濯機や超音波洗浄機などで本器を洗わないでください。

ブザー放音口をふさがないでください。警報音が出なくなります。

電源 ON の状態で電池を外さないでください。

-8-

2 安全上、大切なお知らせ

2-3. 注意事項

2-3. 注意事項
注意












油・薬品などがかかるような場所では使用しないでください。また故意に水中に沈めるようなことは避けてください。
‧ 本器に油・薬品など液体がかかるような場所は避けて使用してください。
‧ ガス吸入口、およびガス排出口は防水構造ではありません。この箇所から雨水等水の浸入がないように注意してください。ガスを検知できなくなります。
‧ 本器を水や泥の溜まるような場所に置かないでください。このような場所に置くとブザー放音口、ガス吸入口などから水や泥が入り故障の原因となることがあります。
‧ 汚水、粉塵、金属粉などを吸引すると、センサの感度が著しく低下します。このような環境下では十分に注意してご使用ください。
温度が-20℃未満または 50℃を超える場所では使用しないでください。
‧ 本器の使用温度範囲は-20℃～50℃です。使用範囲を超えた高温・高湿、高圧、低温環境下でのご使用は避けてください。
‧ 直射日光が当たる場所での長時間にわたる使用は極力避けてください。
‧ 炎天下駐車の車内での保管は避けてください。
本器内に結露が発生しないよう使用範囲を守ってください。
本器中に結露が発生すると、詰まったり、ガスが吸着したりするなど正確なガス検知を行えなくなります
ので、結露の発生は厳禁となります。本器の使用環境と併せて、サンプリング先の温度・湿度には十分注意し、本器中に結露が発生しないようにしてください。必ず使用範囲を守るようにお願いします。
本器の近くでは、トランシーバーを使用しないでください。
‧ 本器の近くでトランシーバーなどによる電波が出力されると、指示に影響する場合があります。トラン
シーバーなどを使用する場合には、本器から離れ、影響の出ないところで使用してください。
‧ 強い電磁波の発生する機器（高周波機器・高電圧機器）の近くでのご使用は避けてください。
ポンプ動作状態表示が回転動作していることを確認して使用してください。
ポンプ動作状態表示が動作していない場合は、正しいガス検知ができません。流量が喪失していないか
確認してください。
動作状態表示が点滅動作していることを確認して使用してください。
動作状態表示が点滅していない場合は、正しいガス検知ができません。

-9-

2 安全上、大切なお知らせ

2-3. 注意事項

注意 センサに関して・センサによっては、検知対象ガス以外のガスに対してもプラス方向に感度を有する場合があります。
それらのガスが共存する環境で使用した場合、実際に存在する検知対象ガス濃度よりも高く指示する
恐れがありますので、注意してください。例えば、シアン化水素（HCN）用の定電位電解式センサでは、
硫化水素（H2S）に対して HCN の約 2 倍の感度を有しますので、H2S が共存する環境下では特に注意が必要です。
センサがプラス方向に感度を有して干渉するガス例
使用するセンサの検知原理（検知対象ガス）／

干渉ガス

定電位電解式（HCN）
定電位電解式（HCN）
定電位電解式（HCN）
定電位電解式（SO2）定電位電解式（SO2）定電位電解式（CO）
定電位電解式（Cl2）
定電位電解式（Cl2）
定電位電解式（PH3）定電位電解式（PH3）定電位電解式（PH3）
接触燃焼式（HC/CH4）
赤外線式（HC/CH4）
PID 式（VOC）

／
／
／
／
／
／
／
／
／
／
／
／
／
／

H2S
SO2
C2H2
H2
CO
H2
SO2
HCｌ
SO2
HCN
H2S
全ての可燃性ガス
炭化水素系の可燃性ガス
全ての VOC・定電位電解式センサは、原理上干渉ガスの種類によって、マイナス方向に感度を有する場合があります。それらのガスが共存する環境では、実際に存在する検知対象ガス濃度よりも低く指示する恐れがありますので、注意してください。
定電位電解式センサがマイナス方向に感度を有して干渉するガス例
使用するセンサの原理（検知対象ガス）／

干渉ガス

定電位電解式（H2S）
定電位電解式（HCN）
定電位電解式（NO2）
定電位電解式（SO2）定電位電解式（NH3）
定電位電解式（PH3）

／
／
／
／
／
／

NO2
NO2
SO2
NO2
H2S
NO2

・シリコーン化合物、ハロゲン化物、硫黄化物などが存在する環境にてニューセラミック式センサを使用した場合、センサの寿命が短くなったり、センサの可燃性ガスに対する感度が低下し、正確な指示を得られなくなる恐れがありますので、注意してください。やむを得ず使用する場合は使用時間を極力短くし、使用後は新鮮な空気を吸引し、指示に異常がないことを確認してください。
・ハロゲン化炭化水素や硫化物が高濃度、または低濃度でも長時間存在する環境にてガルバニ電池式センサを使用した場合、センサの寿命が短くなったり、センサの酸素に対する感度が低下し、正確な指示を得られなくなる恐れがありますので、注意してください。やむを得ず使用する場合は使用時間を極力短くし、使用後は新鮮な空気を吸引し、指示に異常がないことを確認してください。
・本器のニューセラミック式可燃性ガスセンサ＜%LEL＞が正確なガス検知および濃度表示をするには、
ある一定以上の酸素濃度が必要です。
・イナートガス中の酸素濃度を長時間測定する場合は、二酸化炭素濃度は必ず 15%以下の雰囲気で使用してください。15%より高い二酸化炭素濃度の雰囲気で使用する場合は、極力測定する時間を短くしてください。高濃度下で長時間使用すると、酸素センサの寿命が短くなる原因となります。

- 10 -

2 安全上、大切なお知らせ

2-3. 注意事項

注意 センサに関して・高濃度の VOC ガスと接触した場合、CO センサの指示値が上昇する可能性があります。指示値が上昇して戻らない場合、CO センサ用の活性炭フィルタの交換が必要です。活性炭フィルタの交換については
弊社営業部までご連絡ください。
・塩素（Cl2）およびアンモニア（NH3）において、使用温度範囲の下限温度（-20℃）付近では、ガスの性状により応答が遅くなる恐れがありますので注意してください。
・PID 式 VOC センサは、高濃度のメタンガス、エタンガス、プロパンガスなどを吸引した場合、濃度表示部に「－－－－」を表示し、ランプが点滅してブザーが鳴り、一時的に測定できなくなることがあります。これ
らのガスが存在する環境では、濃度表示部に「－－－－」が表示されない場合でも、VOC 濃度を正しく測定できない恐れがありますので注意してください。なお、VOC センサの濃度表示部に「－－－－」が表示されている場合でも、VOC センサ以外の影響を受けないセンサは継続して測定できます。
PID 式 VOC センサの濃度表示部に「－－－－」が表示される干渉ガス例
干渉ガス

／

ガス濃度

メタンガス
エタンガス
プロパンガス／／／

6vol%以上
80vol%以上
90vol%以上

 定期的な点検を必ず行ってください。
本器は安全確保のために定期的な点検を必ず行って使用してください。点検を行わずに使用を続けると、センサの感度が変化し、正確なガス検知を行えません。
 その他‧ むやみにボタンを押すと、各設定が変更されてしまい、警報が正常に作動しないことがあります。本取扱説明書に記載されている以外の操作は行わないでください。
‧ 落下させたり、衝撃を与えたりしないでください。精度の低下を招くことがあります。
‧ 本器を充電しながら使用しないでください。
 ブザーの放音口を先の尖ったもので突かないでください。故障や破損の原因となり、異物などが浸入する可能性があります。
 LCD 表示部のパネルシートを剥がさないでください。防塵性能が損なわれます。
 赤外線ポート部にラベルなどを貼付しないでください。赤外線通信ができなくなります。
 電池交換について‧ 乾電池ユニットの電池を交換する際は、必ず本器の電源を OFF にしてから行ってください。
‧ 乾電池ユニットの電池を交換する際は、3 本とも新しい電池を使用してください。
‧ 電池の極性に注意してください。
 使用に関して‧ 低温度の環境では、電池の性能上、使用時間が短くなります。
‧ 低温時は LCD 表示の応答が遅くなる場合があります。
‧ ゼロ校正は、使用環境に近い状態の圧力、温湿度条件下かつ新鮮な空気中で行ってください。
‧ ゼロ校正は指示が安定してから行ってください。
‧ 保管場所と使用場所の温度が 15℃以上急変するような場合、電源を入れた状態で使用場所と同様の環境下にて 10 分程度馴染ませ、新鮮な大気中でゼロ校正を実施してから使用してください。
‧ 本器の汚れを拭き取る際、水をかけたり、アルコールやベンジンなどの有機溶剤を用いたりしないでください。本器表面が変色したり、損傷したりする可能性があります。
‧ 長期間使用しない場合でも、6 ヶ月に一度は電源を入れ、ポンプが吸引することを確認してください（3 分間程度）。長期間動作させずにいると、ポンプのモーター内のグリスが固まり動作しなくなることがあります。
‧ 長期間使用しない場合は、乾電池を抜いて保管してください。電池の液漏れにより、火災、ケガなどの原因となることがあります。
‧ 長期保管後、再度使用する場合は必ずガス校正を行ってください。ガス校正を含めて、再調整は弊社営業所までご連絡ください。

- 11 -

2 安全上、大切なお知らせ

2-4. セーフティーインフォメーション

2-4. セーフティーインフォメーション
GX-6000 は最大 6 種類のセンサで 6 種類のガスを検知することができます。
4 つのセンサで一般可燃性ガス（%LEL）、酸素（O2）、硫化水素（H2S）および一酸化炭素（CO）を検知します。
残りの 2 つのセンサスロットはセンサ部と基板から構成され、本体とデジタル信号出力でつながるスマートセンサ専用です。スマートセンサは 3 種類の異なる検知原理を採用して複数種類があり、そのうち 2 つを本体に搭載できます。
ガスは内蔵ポンプにて吸引します。
電源ユニットは乾電池ユニット（BUD-6000）またはリチウムイオン電池ユニット（BUL-6000）どちらか選択できます。
電池ユニットはお客様にて交換が可能です。
電池ユニットおよび BUD-6000 の乾電池の交換、また BUL-6000 の充電は、危険場所では行わないでください。
なお、BUL-6000 の充電は、専用充電器（BC-6000）または SDM-6000 にて可能です。
防爆仕様
GX-6000
防爆等級
Ex ia IIB T4 Ga
II1G Ex ia IIB T4 Ga
Ex ia IIC T4 Ga
II1G Ex ia IIC T4 Ga
Ex ia IIB T4 Ga
II1G Ex ia IIB T4 Ga
Ex ia IIC T4 Ga
II1G Ex ia IIC T4 Ga
Ex ia IIB T3 Ga
II1G Ex ia IIB T3 Ga
Ex ia IIC T3 Ga
II1G Ex ia IIC T3 Ga

周囲温度範囲

可燃性ガス
センサ

電池の種類

-20°C to +50°Cあり

BUL-6000-20°C to +50°C

なし

BUL-6000-20°C to +50°Cあり

BUD-6000LR6(TOSHIBA)-20°C to +50°C

なし

BUD-6000LR6(TOSHIBA)-20°C to +50°Cあり

BUD-6000
MN1500(DURACELL)

なし

BUD-6000
MN1500(DURACELL)-20°C to +50°C

・周囲温度範囲（充電時） ： 0°C to +40°C

電気的仕様
・リチウムイオン電池ユニット : BUL-6000
BUL-6000 の電池パックにはマクセル製の INR18650PB1、SDI 製の INR18650-15M、または SONY 製の US18650VT3 リチウムイオン電池セルが 2 つ並行に接続されています。（Um=250V）
・乾電池ユニット : BUD-6000
使用できる乾電池は東芝製(LR6)または DURACELL 製(MN1500)単 3 形アルカリ乾電池です。

- 12 -

2 安全上、大切なお知らせ

2-4. セーフティーインフォメーション

認証番号
・IECEx ： IECEx PRE 15.0011
・ATEX ： Presafe15 ATEX6171X
・UKEX ： DNV 22 UKEX 25912X
適用規格
・IEC 60079-0：2017
・IEC 60079-11：2011

・EN IEC 60079-0:2018
・EN 60079-11:2012

・BS EN IEC60079-0:2018
・BS EN 60079-11:2012

本型式認証取得に係わる条件
ATEX 指令付属書 II 1.5.5 項に基づく測定機能は、本 EU 型式認証書には含まれません。
ガスパフォーマンス及びセーフティデバイスの要求については、別途欧州整合規格の要求に適合する必要があります。
規制 UKSI 2019 No. 1107 の schedule 1 第 24 項に項に基づく測定機能は、本 UK 型式認証書には含まれません。ガスパフォーマンス及びセーフティデバイスの要求については、別途英国指定規格の要求に適合する必要があります。
注意事項
・リチウムイオン電池ユニットを危険場所で充電しないでください。
・リチウムイオン電池ユニットは専用充電器で充電してください。
・電池ユニットを危険場所で交換しないでください。
・危険場所で乾電池を交換しないでください。
・機器を分解または改造しないでください。
・乾電池ユニットには東芝製（LR6）または DURACELL 製(MN1500)単 3 形アルカリ乾電池のみを使用してください。
INST. No. 0 0 0 0 0 0 0 0 0 0 0
AB C
DE
A: 製造年 (0-9)
B: 製造月 (1-9 月、X<10 月>、Y<11 月>、Z<12 月>)
C: 製造ロット
D: シリアルナンバー
E: 工場コードPhone
Fax
E-mail: +81-3-3966-1113: +81-3-3558-9110: intdept@rikenkeiki.co.jp- 13 -

3 製品の構成

3-1. 本体および標準付属品

3
3

製品の構成

3-1. 本体および標準付属品
梱包箱を開けて、本器と付属品を確認してください。
万一、足りないものがありましたら、販売店または最寄りの弊社営業所までご連絡ください。

本体
本器の各部の名称と働きおよび LCD 表示は、「3-2. 各部の名称と働き」（P.17）を参照してください。

GX-6000 本体- 14 -

3 製品の構成

3-1. 本体および標準付属品

標準付属品
リチウムイオン
電池ユニット（BUL-6000）
（※1）
1個

充電器
（※1）
1個

乾電池ユニット（BUD-6000）（※2）
1個

単 3 形アルカリ
乾電池
（※2）
3本

プロテクトカバー1個

ベルトクリップ
1個（ねじ 3 個）

何かにぶつけたときなどの衝撃から、本器を保護します。

本器をベルトにかけられます。

テーパーノズル
1本

ハンドストラップ
1本

LCD 保護フィルム
1枚

CO2 除去フィルタ（CF-284）
1個

細かい傷から画面を守ります。

CO2 センサ搭載の場合のみ付属します。

活性炭フィルタ(CF-8350)
1個

活性炭フィルタ(CF-8501)
1個

VOC センサ搭載の場合のみ付属します。
（CO2 センサも搭載している 仕 様 の 場 合 、 CF-8501が付属されます。）

VOC センサ及びCO2 センサ搭載の場合のみ付属
します。

製品保証書

取扱説明書

※1, 2 リチウムイオン電池ユニットおよび充電器、または乾電池ユニットおよび単 3 形アルカリ乾電池はどちらか一方が付属されます。

- 15 -

3 製品の構成

3-1. 本体および標準付属品

危険防爆に関して
回路・構造などの改造または変更は、行わないでください。

酸素濃度の測定では、空気と可燃性ガスまたは蒸気と毒性ガスの混合物以外の測定をしないでください。

本器を携帯して危険な場所で使用する場合は、静電気の帯電による危険防止総合対策として、① 使用する衣服は帯電防止作業服、履物は導電性履物（帯電防止作業靴）を使用する② 屋内では、導電性作業床（漏洩抵抗 10MΩ以下）の環境で使用するとしてください。

接続できる電池ユニットは、BUL-6000（合格番号 TC21156）または BUD-6000（合格番号 TC21157）です。

本器の定格は以下の通りです。
スマートセンサ 1 回路：許容電圧 4.95V、許容電流 0.770A、許容電力 0.787W
スマートセンサ 2 回路：許容電圧 4.95V、許容電流 0.770A、許容電力 0.787W
メイン回路：許容電圧 4.95V、許容電流 1.112A、許容電力 1.137W
ポンプ回路：許容電圧 4.95V、許容電流 0.770A、許容電力 0.787W
モータ回路：許容電圧 4.95V、許容電流 0.209A、許容電力 0.214W
ブザー回路：許容電圧 4.95V、許容電流 0.355A、許容電力 0.363W

注記 充電器は、DIN レールに取り付けて利用できます。
DIN レールは、IEC715 トップハット型 TH35 を使用してください。
取り付ける場合は、DIN レールの返りの部分を本体の爪に引っ掛け、
DIN レールの返りの部分にストッパを引っ掛けます。
 外す場合は、ストッパを下方向に押します。

爪

ストッパ

- 16 -

3 製品の構成

3-1. 本体および標準付属品

オプション品（別売）
リチウムイオン
電池ユニット（BUL-6000）
1個

充電器
1個

乾電池ユニット（BUD-6000）
1個

単 3 形アルカリ
乾電池
3本

ガス採集棒
1本

プレフィルターチューブ(CF-8338)
1 箱（10 本入り）

各種ガス採集チューブ（0.75m/5m/10m/20m/30m）

チューブホルダー(GF-284)
VOC<10.0eV >センサを搭載した仕様のみ使用可能

VOC<10.0eV>
センサを搭載した仕様のみ使用可能

各種フィルタ

データロガマネジメント
プログラム

PID リスト設定
プログラム

注意


特定の毒性ガスや溶剤系 VOC など、吸着性の高いガスを測定する際にオプションのガス採集チューブ（0.75m/5m/10m/20m/30m）を使用した場合、チューブ内にガスが吸着し、測定ポイントに実際に存在する検知対象ガス濃度よりも低く指示する恐れがありますので、注意してください。

- 17 -

3 製品の構成

3-2. 各部の名称と働き

3-2. 各部の名称と働き本体、電池ユニットの各部の名称と働きおよび LCD 表示について記載しています。

本体

名称

主な機能

(1)

LCD 表示部

ガス濃度などを表示します。

(2)

ブザー放音口

操作音や判定音を放出する口です。（ふさがないでください。）

(3)

警報窓

警報時に、ランプが赤く点滅します。

(4)

照明ライト

（照明ライト）ボタンの長押しで点灯します。

(5)

赤外線通信ポートデータロガマネジメントプログラム使用時に、パソコンとデータ通信を
行います。

(6)

▲/AIR ボタン

検知画面では、ゼロ校正を行います。また DISP モード、ユーザー
モード時にはカーソル（＞）を上に移動します。

(7)

SHIFT/▼/（PANIC）ボタンDISP モード、ユーザーモード時にはカーソル（＞）を下に移動します。
緊急時には、長押しでパニック警報を発報します。

DISP/LOCK ボタン

DISP モードを表示して、切り替えます。
また、表示上下反転（P.71）を設定している場合に長押しすると、表示を固定します。

(8)- 18 -

3 製品の構成

3-2. 各部の名称と働き
名称

主な機能

(9)

RESET/ （照明ライト）ボタン

警報を確認、解除します。また長押しで上部の照明ライトが点灯しま
す。

(10)

POWER/ENTER ボタン

電源を入/切します。また DISP モード、ユーザーモード時には選択項目を確定します。

(11)

ガス吸入口

ガスを吸引する口です。（ふさがないでください。）

(12)

ガス排出口

吸引したガスを排出する口です。（ふさがないでください。）(13)

ストラップ用穴（2 箇所）

付属のハンドストラップを取り付けます。

(14)

センサカバー

内部のセンサを保護しています。交換時以外は開けないでください。

(15)

フィルタケース

内部のダストフィルタを保護しています。点検・交換時以外は外さな
いでください。

(16)

電池ユニット脱着用レバー

スライドさせながら押すと、電池ユニットを外すことができます。

(17)

電池ユニット接続端子

本器に電池ユニットの電源を供給する端子です。

注意




ブザー放音口を、先の尖ったもので突かないでください。水や異物などが侵入して、故障や破損の原因となります。
表面のパネルシートを剥がさないでください。防水・防塵性能が損なわれます。
赤外線通信ポートにラベルなどを貼り付けないでください。赤外線通信ができなくなります。

注記
 本取扱説明書では、複数の機能名が割り付けられたボタンは、操作説明で次のような名称としています。
例）名称「POWER/ENTER ボタン」を、・電源入/切の操作時には POWER ボタン
・設定時の確定ボタンとして操作には、ENTER ボタン
としています。

- 19 -

3 製品の構成

3-2. 各部の名称と働き電池ユニット＜リチウムイオン電池ユニット（BUL-6000）＞＜乾電池ユニット（BUD-6000）＞- 20 -

3 製品の構成

3-2. 各部の名称と働き

LCD 表示部

名称
動作状態表示電池残量表示
時計表示
ポンプ動作状態表示
可燃性ガス 濃度表示
酸素 濃度表示硫化水素 濃度表示
一酸化炭素 濃度表示
スマートセンサの(9)
検知対象ガス濃度表示（10） バー表示
(1)
(2)
(3)
(4)
(5)
(6)
(7)
(8)

主な機能
動作状態を表示します。正常時は点滅します。
電池残量を表示します。電池残量の目安は注記を参照してください。
時刻を表示します。
吸引状態を表示します。正常時は回転します。

ガス濃度を数字で表示します。

ガス濃度をバーで表示します。

注記 ガス濃度の表示位置は入れ替えることができます。入れ替える方法については「ガス濃度値の表示位置を入れ替える」（P.76）を参照してください。
 電池残量の目安を以下のように表示します。
十分に残っています。
少なくなっています。
充電（電池交換）してください。
さらに電池残量が少なくなると、電池マークが点滅します。

- 21 -

4 警報動作

4-1. ガス警報動作

4
4

警報動作

4-1. ガス警報動作＜ガス警報の種類＞「ガス警報」は、検知したガス濃度が下表に示す警報設定値に達する、または超えたときに発報します。（自己保持動作）
ガス警報の種類は、第一警報（AL1）、第二警報（AL2）、TWA 警報、STEL 警報、OVER 警報（オーバースケール）
です。
警報の種類
第一警報
第二警報
TWA 警報STEL 警報OVER 警報
酸素（O2）

19.5vol%

18.0vol%－－

40.0vol%

可燃性ガス（HC/CH4）＜%LEL＞

10%LEL

50%LEL－－

100%LEL

硫化水素（H2S）

1.0ppm10.0ppm

1.0ppm5.0pm

30.0ppm

一酸化炭素（CO）

25ppm

50ppm

25ppm

200ppm500ppm

揮発性有機化合物（VOC）＜10.6eV / ppb＞

5000ppb

10000ppb--

40000ppb

揮発性有機化合物（VOC）＜10.6eV / ppm＞

400.0ppm

1000ppm--

4000ppm

揮発性有機化合物（VOC）＜10.0eV / ppm＞※

5ppm

10ppm

-

-

100ppm

二酸化硫黄（SO2）2.00ppm

5.00ppm

2.00ppm

5.00ppm

99.90ppm

二酸化窒素（NO2）3.00ppm

6.00ppm

3.00ppm－

20.00ppm

シアン化水素（HCN）

5.0ppm10.0ppm－

4.7ppm

15.0ppm

アンモニア（NH3）25.0ppm

50.0ppm

25.0ppm

35.0ppm

400.0ppm

塩素（Cl2）0.50ppm

1.00ppm

0.50ppm

1.00ppm

10.00ppm

ホスフィン（PH3）0.30ppm

1.00ppm

0.30ppm

1.00ppm

20.00ppm

二酸化炭素（CO2）＜vol%＞0.50vol%

3.00vol%

0.50vol%

3.00vol%

10.00vol%二酸化炭素（CO2）＜ppm＞

5000ppm-

5000ppm-

10000ppm

可燃性ガス（HC）＜%LEL / vol%＞10%LEL/-

50%LEL/--/--/-30.0vol%

可燃性ガス（CH4）＜%LEL / vol%＞10%LEL/-

50%LEL/--/--/-100.0vol%※ノーマルモードでの警報点です。ベンゼンセレクトモードではガス警報は発報しません。

- 22 -

4 警報動作

4-1. ガス警報動作＜ガス警報のブザー鳴動とランプ点滅動作＞
ガス警報の動作は、ブザーの鳴動、警報窓のランプ点滅、バイブレーション動作を 2 段階で知らせます。
以下に種類ごとの動作を示します。
警報の種類
第一警報
第二警報
TWA 警報STEL 警報OVER 警報
ブザー鳴動

約 1 秒周期の強弱鳴動を繰り返す。
「ピーピー」

約 0.5 秒周期の強弱鳴動を繰り返す。
「ピーピーピーピー」

約 1 秒周期の強弱鳴動を繰り返す。
「ピーピー」

約 1 秒周期の強弱鳴動を繰り返す。
「ピーピー」

約 0.5 秒周期の強弱鳴動を繰り返す。
「ピーピーピー
ピー」

警報窓のラ
ンプ点滅

約 1 秒周期の点
滅動作を繰り返す。

約 0.5 秒周期の点滅動作を繰り返す。

約 1 秒周期の点
滅動作を繰り返す。

約 1 秒周期の点
滅動作を繰り返す。

約 0.5 秒周期の点滅動作を繰り
返す。

バイブレー
ション動作

警報時に振動『警報パターン（H-HH）』『警報パターン（L-LL）』：1 秒周期：0.5 秒周期：1 秒周期：0.5 秒周期
RESET

RESET

濃度

濃度

第二警報設定値
第一警報設定値

第一警報設定値
第二警報設定値

0

0

時間

時間
警報ランプ
ブザー

警報ランプ
ブザー

＜ガス警報の表示動作＞
ガス警報では当該ガス濃度表示および内容表示を交互に表示します。
検知範囲を超えると（オーバースケール）、ガス濃度表示部に「OVER」と表示します。

表示例
メタン（CH4）濃度：10%LEL
第一警報発報
警報の種類

第一警報

第二警報

TWA 警報STEL 警報OVER 警報

LCD 表示ガス濃度値と「 AL1 」 を 交 互 に表示する。

ガス濃度値と「 AL2 」 を 交 互 に表示する。

ガス濃度値と「TWA」を交互に表示する。

ガス濃度値と「STEL」を交互に表示する。

ガス濃度表示部に 「 OVER 」 を表示し、「OVER」と交互に表示する。

- 23 -

4 警報動作

4-1. ガス警報動作

警告

ガス警報が発報された場合は大変危険です。お客様の判断により適切な処置を行ってください。

注記 警報時の動作は、DISP モードの警報テスト（P.64）で確認することができます。ただし、警報テストでは、表示は切り替わりません。

- 24 -

4 警報動作

4-2. 故障警報動作

4-2. 故障警報動作「故障警報」は、本器内での異常動作を検知して故障警報として発報します。（自己保持動作）
警報の種類は、システム異常、電池電圧異常、時計異常、流量低下異常、センサ異常、校正不良です。
警報時の動作は、ブザー鳴動、警報窓のランプ点滅で知らせます。
・ ブザー鳴動：約 1 秒周期の間欠鳴動を繰り返す。「ピーピー、ピーピー」
・ 警報窓のランプ点滅：約 1 秒周期の点滅動作を繰り返す。
故障警報時の表示例を以下に示します。

システム異常電池電圧異常流量低下異常センサ異常／校正不良

時計異常

故障警報の発報時には、原因を究明し適切な対処を行ってください。
機器に問題があり、故障が頻発する場合は、速やかに弊社営業所にご連絡くださいますようお願いします。

注記 故障内容（エラーメッセージ）の詳細については「トラブルシューティング」（P.89）を参照してください。

- 25 -

4 警報動作

4-3. パニック警報

4-3. パニック警報
パニック警報は、使用者が周囲に異常を知らせるために、手動で警報を発報する機能です。

警告

パニック警報は、使用者または周囲の人の判断を補助するものです。その検知結果により生命・安全の保障をするものではありません。本機能だけに頼った運用は、行わないでください。
パニック警報を発報する場合は、周囲の状況を確認し、適切に使用してください。

パニック警報のブザー鳴動とランプ点滅動作
警報の種類
ブザー鳴動
警報窓の
ランプ点滅

予備警報

本警報

約 0.5 秒周期の断続鳴動を繰り返す。
「ピッ、ピッ、ピッ、ピッ」

約 1 秒周期の強弱鳴動を繰り返す。
「ピーピーピーピー」

約 0.5 秒周期の点滅動作を繰り返す。

約 1 秒周期の点滅動作を繰り返す。

パニック警報の発報と警報パターン
使用者が異常を感じ、パニック警報を発報する場合、PANIC ボタンを長押しします。
パニック警報の場合、5 秒間の予備警報の後、本警報に切り替わります。

注記
パニック警報の予備警報および本警報を止めるときは、RESET ボタンを押してください。

- 26 -

4 警報動作

4-4. マンダウン警報

4-4. マンダウン警報
マンダウン警報は、内蔵のモーションセンサにより、本器を装着した使用者の動きを検知し、一定時間、使用者に動きがなかった場合に発報する警報機能のことです。
マンダウン警報は、通常 OFF 設定となっており使用できません。本機能をご利用になる場合は、販売店または最寄りの弊社営業所までご連絡ください。

警告

マンダウン警報は、周囲の人の判断を補助するものです。その検知結果により生命・安全の保障をする
ものではありません。本機能だけに頼った運用は、行わないでください。
マンダウン警報を発報する場合は、周囲の状況を確認し、適切に使用してください。

マンダウン警報のブザー鳴動とランプ点滅動作
警報の種類

予備警報 1

予備警報 2

本警報

ブザー鳴動

約 1 秒周期の断続鳴動を繰り返す。
「ピッ、ピッ」

約 0.5 秒周期の断続鳴動を繰り返す。
「ピッ、ピッ、ピッ、ピッ」

約 1 秒周期の強弱鳴動を繰り返す。
「ピーピーピーピー」

警報窓の
ランプ点滅

約 1 秒周期の点滅動作を繰り返す。

約 0.5 秒周期の点滅動作を繰り返す。

約 1 秒周期の点滅動作を繰り返す。

マンダウン警報の表示と警報パターン
使用者の動きに異常を検知すると、ランプが点滅し、振動しながら予備警報 1、
予備警報 2、本警報と段階的に発報します。
本警報を発報すると、LCD 表示部の時計表示が「MAN DOWN」になります。
予備警報から本警報へと切り替わる時間は、下記の通りです。
・予備警報 1：検知から 60 秒後・予備警報 2：検知から 75 秒後・本警報：検知から 90 秒後

注記 マンダウン警報の予備警報は、使用者の動きを検知すると、発報を止めて測定状態に戻ります。
 マンダウン警報の本警報を止めるときは、RESET ボタンを押してください。

- 27 -

5 使用方法

5-1. ご使用するにあたって

5
5

使用方法

5-1. ご使用するにあたって
本器を初めてご使用になる方、および既にご使用経験のある方も、使用方法の注意事項を必ず守ってください。
これらの注意事項を守らない場合には、機器の故障が生じ、正常にガスを検知できない場合があります。

5-2. 始動準備
警告
 本器の表示部には、傷防止のため出荷時に保護フィルムが貼付されています。
ご使用になる前に、必ずこの保護フィルムを剥がしてください。この保護フィルムを貼付したままの製品
では、防爆性能を担保できません。

ガス検知を開始する前に、以下の内容を確認してください。
・傷防止のため出荷時に表示部に貼付された保護フィルムが付いていないこと
・電池残量が十分であること・テーパーノズルが折れていないこと、穴が開いていないこと
・本器内のフィルタが汚れていないこと、目詰まりがないこと
・本体とテーパーノズルが正しく接続されていること

5-2-1. リチウムイオン電池ユニット（BUL-6000）の充電と装着
本器を初めて使用する場合や、リチウムイオン電池ユニット内の充電池の電池残量が少ない場合は、以下の手順で、付属の充電器で充電してください。

危険





リチウムイオン電池ユニットの交換は、安全な場所で行ってください。
充電は、安全な場所で、付属の充電器を使用して行ってください。
充電は、0℃～＋40℃の環境下で行ってください。
本ユニットの定格は、以下の通りです。
最大電圧：4.2V、周囲温度：－20℃～＋50℃- 28 -

5 使用方法

5-2. 始動準備

注意








電池ユニットの交換は、必ず本器の電源を OFF にしてから行ってください。
充電を行いながら、本器を使用しないでください。正しい測定ができません。また、電池寿命が短くなるな
ど、充電池の劣化が早まります。
本体が濡れている状態で充電をしないでください。充電器は防水・防塵構造ではありません。
充電器は防爆仕様ではありません。
リチウムイオン電池ユニットを装着した後は、電池蓋を完全にロックしてください。電池蓋のロックが不完
全な場合は、電池ユニットが脱落したり、すき間から水が浸入する可能性があります。
ゴムパッキンを傷付けないでください。
防水／防塵性能を維持するために、異常の有無に関わらず、ゴムパッキンは 2 年ごとに交換することをお勧めします。

＜リチウムイオン電池を充電する＞
1

AC アダプタの DC プラグを充電器の DC ジャックに差し込む
DC プラグのコードは、充電器底面の溝から側面にはめ込んでください。

2

AC アダプタをコンセントに差し込む

3

本体を充電器の上からまっすぐ挿入する
充電器が接続されると、充電確認用ランプが赤く点灯します。（最長約 3 時間
で満充電）
充電が終了すると、充電確認用ランプが消灯します。

4

充電が終了したら、AC プラグをコ
ンセントから抜く- 29 -

5 使用方法

5-2. 始動準備＜リチウムイオン電池ユニットの外しかた／取り付けかた＞
1

本体の電源が切れていることを確認する
電源が入っている場合は、
POWER/ENTER ボタンを押して、電源を切ります。

2

電池ユニット脱着用レバーを右方向に力を加えながら押し込む

電池ユニット脱着用レバーを右方向に力を加えながら

3

リチウムイオン電池ユニットを本体
から外す

注意


使用しないときは、必ず AC プラグをコンセントから抜いてください。

注記 電池ユニットを取り付ける場合は、電池ユニット脱着用レバーがロックされていることを確認してください。
 ロックが不完全な場合、電池ユニットが脱着したり、隙間から水が浸入する可能性があります。また、間に微細な異物が挟まっている場
合、水が浸入する可能性があります。
 充電中にリチウムイオン電池ユニットが熱くなることがありますが、異常ではありません。
 充電をすると本体温度が上がります。充電完了後、10 分以上経ってから使用してください。本体が熱いまま使用すると、正しい測定ができない場合があります。
 満充電の状態では、再度充電を行っても、充電できません。
 リチウムイオン電池ユニットは、本体から外して単体でも充電できます。

- 30 -

5 使用方法

5-2. 始動準備

5-2-2. 乾電池ユニット（BUD-6000）の装着
乾電池ユニットをリチウムイオン電池ユニットの代わりに装着すると、本器を単 3 形アルカリ乾電池 3 本で使用できます。
乾電池ユニットを初めて使用する場合や電池残量が少ない場合は、以下の手順で新品の単 3 形アルカリ乾電池を装着または交換してください。

危険




乾電池ユニットの交換は、安全な場所で行ってください。
電池の交換は、安全な場所で行ってください。
本ユニットの定格は、以下の通りです。
最大電圧：4.95V、電源：LR6（株式会社東芝製 DC1.5V）3 本、周囲温度：－20℃～＋50℃

注意







電池ユニットの交換は、必ず本器の電源を OFF にしてから行ってください。
電池を交換する際は、必ず本器の電源を切ってから行ってください。
電池を交換する際は、必ず爆発性ガスのない安全な場所で行ってください。
電池を交換する際は、3 本とも新しい電池を使用してください。
電池を装着する際は、極性に注意してください。
電池を装着した後は、電池蓋を完全にロックしてください。電池蓋のロックが不完全な場合は、乾電池が脱落したり、すき間から水が浸入する可能性があります。また、電池蓋の間に微細な異物が挟まると、水が浸入する可能性があります。

＜乾電池ユニットの外しかた／取り付けかた＞
1

本体の電源が切れていることを確認する
電源が入っている場合は、
POWER/ENTER ボタンを押して、電源を切ります。

2

電池ユニット脱着用レバーを右方向に力を加えながら押し込む

電池ユニット脱着用レバーを右方向に力を加えながら

3

乾電池ユニットを本体から外す- 31 -

5 使用方法

5-2. 始動準備

注記
 電池ユニットを取り付ける場合は、電池ユニット脱着用レバーがロックされていることを確認してください。
 ロックが不完全な場合、電池ユニットが脱着したり、隙間から水が浸入する可能性があります。また、間に微細な異物が挟まっている場合、水が浸入する可能性があります。

＜乾電池を交換する＞
1

電池蓋のロックプレートを外す

2

電池蓋を開ける

3

電池の極性に注意して、新しい電池を入れる
古い電池が入っている場合は、取り外します。

4

電池蓋を閉じ、ロックプレートを閉める
ロックプレートは、「カチッ」と音がするまでしっかりと閉めます。

- 32 -

5 使用方法

5-2. 始動準備

5-2-3. テーパーノズルの装着測定する際には、本器のガス吸入口に、テーパーノズルを装着してください。

危険

当社指定以外のテーパーノズルおよびその他部品を、本器に使用しないでください。

5-2-4. プレフィルターチューブ（CF-8338）及びチューブホルダー （GF-284）の装着（VOC＜10.0eV＞センサを搭載している仕様のみ）
VOC＜10.0eV＞センサを搭載している場合、ベンゼンセレクトモードにてベンゼンに対する選択性を高めて濃度を測定することができます。ベンゼンセレクトモードでは、ベンゼン測定用プレフィルターチューブ（CF-8338）及び
チューブホルダー（GF-284）を装着して測定を行います。プレフィルターチューブチューブ及びチューブホルダーは、
下記の手順で GX-6000 に装着してください。

注意
プレフィルター（CF-8338）に同梱された取扱説明書をよく読んでから取り扱ってください。

1

プレフィルターチューブの両端を、チューブホルダーのチューブカッ
ターでカットする
プレフィルターチューブの先端をチップカッターの開口部に入れて、プレフィ
ルターチューブを一回転させます。
プレフィルターチューブの根本近くを持って手前に傾け、先端を折ります。

2

プレフィルター

チューブカッター(ローレットねじ)
チューブホルダー

チューブホルダーのキャップ部にある爪を押しながらキャップ部を引っぱり、取り外す

キャップ部
爪- 33 -

5 使用方法

3

キャップ部にプレフィルターチュー
ブを挿入する5-2. 始動準備＜正しい挿入例＞
斜線部が見えない＜誤った挿入例＞
斜線部が見えるプレフィルターチューブに貼ってあるラベルの 斜線部が隠れるくらいまで、挿入して ください。

斜線部

4

チューブホルダーにキャップ部を装着する
キャップ部は、「カチッ」と音がするまでしっかりと差し込みます。

5

チューブホルダーを GX-6000 に装着する
テーパーノズル、チューブホルダー、GX-6000 の順で装着します。

- 34 -

5 使用方法

5-2. 始動準備注意

低温環境ではチューブホルダーのパッキンが固くなり、プレフィルターを装着しにくくなります。室温環境にてプレフィルターをチューブホルダーに装着し、低温環境ではなるべく短時間で使用してください。
プレフィルター用いて測定を行う前に、校正コードを設定する必要があります。設定方法については「プレフィルターの校正コード入力」（P.44）を参照してください。

注記 オプションのガス採集チューブを使用する場合は、テーパーノズル、チューブホルダー、ガス採集チュー
ブ、GX-6000 の順で接続してください。

チューブカッターの取り外し
チューブカッターを用いてカットしたガラス片は、チューブカッターを取り外して捨てることができます。
定期的に取り外してガラス片を捨ててください。

ゴムキャップ

チューブカッター(ローレットねじ)

チューブホルダー

チューブカッター(ローレットねじ)を回して固定(ねじ固定)

中のガラス片が出てしまう可能性がある為
ゴムキャップは奥までしっかりとチューブカッターにかぶせてください。
- 35 -

5 使用方法

5-3. 始動方法

5-3. 始動方法
電源を入れると、日時や警報点などの各種設定を表示し、ノーマルモードの測定画面を表示します。

電源を入れる
POWER/ENTER ボタンをブザーが “ピッ” と鳴るまで（5 秒以上）押す
電源が入ります。

LCD 表示部が全点灯します。

- 36 -

5 使用方法

5-3. 始動方法

電源投入から測定画面までの遷移
電源投入後、以下のように LCD 表示が自動的に切り替わり、測定画面になります。

10.0eV

10.6eV

ブザーが “ピッピッ” と
2 回鳴り、測定画面となります。

- 37 -

5 使用方法

5-3. 始動方法

注意

始動後、ガス検知を行う前に「5-4. ゼロ校正」（P.39）を実施してください。

注記 センサに異常があった場合、測定画面になる直前に測定値に「FAIL」が表示され、センサ異常警報を発報します。このようなときは RESET ボタンを押して、一時的にセンサ異常警報を解除してください。ただし、全てのセンサに異常があった場合は警報解除できません。警報解除後は、センサに異常のあったガス濃度
表示部に「－－－－」と表示され、センサに異常のあったガスの検知ができません。速やかに販売店または最寄りの弊社営業所までご連絡ください。
 内蔵時計に異常があった場合、故障警報 FAIL CLOCK を発報することがあります。このようなときは
RESET ボタンを押してください。一時的に故障警報が解除され、時計の日時はずれたまま測定を開始します。
暖機
ウォームアップ画面が表示されます。
日時
年月日と時刻が表示されます。日付／時刻設定および表示形式は、「6-3. ユーザーモード設定」（P.75）で設定できます。
電池
 画面上に電池残量（電圧）を表示します。
 画面中央に使用しているバッテリ（リチウムイオン電池、または乾電池）を表示します。
 画面下にガス警報動作の設定（LATCHING＜自己保持＞）を表示します。
GAS
検知対象ガス名を表示します。また、スマートセンサを搭載している場合、検知原理を記号で以下のように表示します。
記号
検知対象ガス
検知原理Ⓟ
揮発性有機化合物（VOC）
光イオン化式（PID 式）
二酸化硫黄（SO2）
二酸化窒素（NO2）
シアン化水素（HCN）Ⓔ
定電位電解式
アンモニア（NH3）
塩素（Cl2）ホスフィン（PH3）二酸化炭素（CO2）＜vol%＞二酸化炭素（CO2）＜ppm＞Ⓓ
赤外線式
可燃性ガス（HC）＜%LEL / vol%＞可燃性ガス（CH4）＜%LEL / vol%＞GAS スマート 1／GAS スマート 2
揮発性有機化合物（VOC）を検知対象とする仕様において、イソブチレンまたは読み替え設定されたガス名を表示します。読み替え設定については、「PID ガス選択設定」（P.61）を参照してください。
F.S.
検知対象ガスのフルスケール値を表示します。
AL1
検知対象ガスの 1 段階目の警報設定値を表示します。
AL2
検知対象ガスの 2 段階目の警報設定値を表示します。
STEL
検知対象ガスの STEL の警報設定値を表示します。STEL 値は、使用者が 15 分間連続被曝した場合で、さらに毎日の被曝が TWA 値以下であれば健康上使用者に悪影響を及ぼさない有害物質の濃度です。
TWA
検知対象ガスの TWA の警報設定値を表示します。TWA 値は、1 日 8 時間、または週 40 時間の平常作業において反復曝露しても、ほとんど全ての使用者が健康上悪影響をこうむることがないと考えられる有害物質の時間荷重平均値のことです。
- 38 -

5 使用方法

5-4. ゼロ校正

5-4. ゼロ校正
ゼロ校正は、現在のガス濃度を正確に測定するためのゼロ調整です。

注意

始動後、ガス検知を行う前にゼロ校正を実施してください。

CO2 除去フィルタ（CF-284）の装着（CO2 センサを搭載している仕様のみ）
CO2 を検知対象とする仕様の場合、ゼロ校正をする際に CO2 除去フィルタ（CF-284）を使用して、大気中の CO2を除去する必要があります。

警告

VOC と CO2 を検知対象とする仕様の場合は、ゼロ校正時に CO2 除去フィルタに加えて活性炭フィルタを使用する必要があります。その場合、本器、活性炭フィルタ、CO2 除去フィルタの順でフィルタを装着してください。

CO2 除去フィルタは、グレーのチューブを外し、側面に記載された矢印（→）が本体のガス吸入口に向くように取り付けてください。なお、使用するときには、CO2 除去フィルタを立ててください。横にしたまま吸引すると、空気中のCO2 を吸収できない場合があります。

矢印（→）をガス吸入口に向けて取り付けてください。

CO2 除去フィルタ 正しい使いかた

CO2 除去フィルタ 誤った使いかた- 39 -

5 使用方法

5-4. ゼロ校正

1 本で使用可能な回数は、大気中の二酸化炭素濃度によって異なります。また、CO2 除去フィルタの気密の度合
い、保管温度、湿度によっても異なります。
1 回 1 分間の吸引で、およそ下表の回数が目安となりますが、環境中の二酸化炭素濃度が特定できない場合は、余裕をもって使用してください。
測定環境中の二酸化炭素（CO2）濃度

保管状況を勘案した推定使用可能回数

500ppm

約 1000 回

1000ppm

約 500 回

2000ppm

約 200 回

4000ppm

約 100 回注意
 使用するときには、CO2 除去フィルタを立ててください。横にしたまま吸引すると、空気中の CO2 を吸収できない場合があります。
 フィルタを装着した状態で新鮮な空気を 1 分間程度吸引した後、ゼロ校正を実施してください。
 ゼロ校正中に、高濃度の二酸化炭素を吸引しないでください。
 ゼロ校正中に、吸引口に呼気を吹きかけないでください。
 CO2 除去フィルタの使用後は、大気との通気を遮断してください。吸収剤は、大気が混入すると、大気中の二酸化炭素を吸収して吸収性能が下がります。
 CO2 除去フィルタは、直射日光の当たらない乾燥した場所に保管してください。

注記
 １本で使用可能な回数は、大気中の二酸化炭素濃度によって異なります。また、CO2 除去フィルタの気密の度合い、保管温度、湿度によっても異なります。
 １回１分間の吸引でおよそ上記の表の回数が目安となりますが、環境中の二酸化炭素濃度が特定できな
い場合は余裕を見て使用してください。

活性炭フィルタの装着（VOC センサを搭載している仕様のみ）
VOC を検知対象とする仕様の場合、ゼロ校正をする際に活性炭フィルタを使用して、大気中の VOC を除去する必要があります。
使用する活性炭フィルタは、CO2 センサを同時に搭載するか否かで種類が異なります。
CO2 センサの有無

使用する活性炭フィルタの型式無し

CF-8350有り

CF-8501- 40 -

活性炭フィルタの外観

5 使用方法

5-4. ゼロ校正

活性炭フィルタは、両側のキャップを外し、側面に記載された矢印（→）が本体のガス吸入口に向くように取り付けてください。

例 CF-8350
矢印（→）を
ガス吸入口に向けて取り付けてください。

警告VOC と CO2 を検知対象とする仕様の場合は、ゼロ校正時に活性炭フィルタに加えて CO2 除去フィル
タを使用する必要があります。その場合、本器、活性炭フィルタ、CO2 除去フィルタの順でフィルタを装着してください。

注意

CO2 センサを搭載している製品では、VOC 除去フィルタを装着すると、一時的に指示が上昇する可能性があります。フィルタを装着した状態で新鮮な空気を 2 分間程度吸引した後、ゼロ校正を実施してください。
活性炭フィルタの使用後は、キャップを取り付けて大気との通気を遮断してください。

- 41 -

5 使用方法

5-4. ゼロ校正

ゼロ校正手順1

測定画面で、AIR ボタンを長押しする

ゼロ校正画面が表示されます。

右記の画面が表示されている間は、AIR ボタンを押し続けてください。
画面が表示される前に指を離すと、ゼロ校正は行われません。

2

右記の画面が表示されたら、AIR ボタンを離す

ゼロ校正が終わると、右記画面が表示されます。

- 42 -

5 使用方法

5-4. ゼロ校正

ゼロ校正が正常に行われると、自動で測定画面に戻り
ます。

警告


ゼロ校正を周辺空気で行う場合は、周辺が新鮮な大気であることを確認してから行ってください。雑ガスなどが存在する状態で行うと、正しいゼロ校正が行えず、実際にガスが漏洩した場合に危険です。

VOC を検知対象とする仕様の場合は、活性炭フィルタ（CF-8350 または CF-8501）を装着してゼロ校正を行ってください。

CO2 を検知対象とする仕様の場合は、CO2 除去フィルタ（CF-284）を装着してゼロ校正を行ってください。
 VOC と CO2 を検知対象とする仕様の場合は、ゼロ校正時に活性炭フィルタおよび CO2 除去フィルタを使用する必要があります。その場合、本器、活性炭フィルタ、CO2 除去フィルタの順でフィルタを装着してください。

注意

ゼロ校正は、使用環境に近い状態の圧力、温湿度条件下で、かつ新鮮な空気中で行ってください。
ゼロ校正は、指示が安定してから行ってください。
保管場所と使用場所の温度が 15℃以上急変するような場合は、電源を入れた状態で使用場所と同様の環境下で 10 分程度馴染ませ、新鮮な大気中でゼロ校正を行ってから使用してください。

注記
 ゼロ校正に失敗した場合、「SENSOR」の表示とともに、不良となったセンサの濃度表示部に「FAIL」と表示されます。RESET ボタンを押して故障警報（校正不良）を解除してください。警報が解除されると、校正前の値を表示します。

- 43 -

5 使用方法

5-5. プレフィルターチューブの校正コード入力

5-5. プレフィルターチューブの校正コード入力（VOC<10.0eV>センサを搭載している仕様のみ）
プレフィルター（CF-8338）を用いて測定を行う場合、プレフィルターの梱包箱に記載された校正コードを GX-6000に入力する必要があります。

PID-Pre-Filter
Tube
Benzene(10 Tubes)
ベンゼン（10 回分）Store in a cool and dark place.
冷暗所保管（0～25℃）
Expires End of Nov, 2019
有効期限 2019 年 11 月
CAL CODE ：B
校正コード：B

校正コード 記載例：B

MODEL CF-8338
Lot No.17Y001
Made in Japan1

ノーマルモードの測定画面を表示した状態で、DISP ボタンと SHIFT ボタンを押す

2

プレフィルターの梱包箱に記載された校正コードを入力する

TUBE

ﾍﾞﾝｾﾞﾝ測定
さ▲ ボタンと ▼ ボタンで入力値を変更できます。

⼊⼒
校正ｺｰﾄﾞ︓A- 44 -

5 使用方法

3（VOC<10.0eV>センサを搭載している仕様のみ）

ENTER ボタンで確定する

TUBE

入力確定後、自動で測定画面に戻ります。

ﾍﾞﾝｾﾞﾝ測定
さ終了- 45 -

5 使用方法

5-6. 検知する

5-6. 検知する危険

マンホールの中や密閉された場所を測定する場合には、絶対にマンホールの入り口に身を乗りだした
り、中をのぞき込まないでください。酸素欠乏空気、その他のガスが吹き出す可能性があり危険です。
本器のガス排気口は、酸素欠乏空気などが排出される場合があります。絶対に吸気しないでください。
本器のガス排気口は、高濃度（100%LEL 以上）のガスが排出される場合があります。絶対に火気を近づけないでください。

警告








本器は大気圧状態の雰囲気のガスを吸引するように作られています。本器のガス吸入口、排出口に過大な圧力をかけると、内部から検知ガスが漏洩する可能性があり危険です。過大な圧力がかからないよ
うにして使用してください。
大気圧以上の圧力がかかる検知箇所に、テーパーノズルを直接接続しないでください。内部の配管系統が破損する可能性があります。
ゼロ校正を周辺空気で行う場合は、周辺が新鮮な大気であることを確認してから行ってください。雑ガスなどが存在する状態で行うと、正しい校正が行えず、実際にガスが漏洩した場合に危険です。
ガス警報が発報された場合は大変危険です。お客様の判断により適切な処置を行ってください。
電池電圧が低下すると、ガス検知を行えなくなります。使用中に電池電圧低下警報が発報された場合
は、電源を切り、安全な場所で速やかに充電、または乾電池の交換をしてください。
ブザー放音口をふさがないでください。警報音が聞こえなくなります。

- 46 -

5 使用方法

5-6. 検知する

5-6-1. 基本動作フロー
電源投入後、ノーマルモードの測定画面に進みます。
＜DISP モード＞
TUBE
ﾍﾞﾝｾﾞﾝ測定＜ノーマルモード＞
測定画面

進む︓ENTER
戻る︓DISP
測定︓RESET＜スナップロガ＞

注記
 VOC＜10.0eV＞センサを搭載している仕様の場合、DISP モードよりベンゼンセレクトモードに進みます。

- 47 -

5 使用方法

5-6. 検知する

5-6-2．ノーマルモード
ノーマルモードで、テーパーノズルを検知する場所に近づけ、LCD 表示部の数値を読み取ってください。

スマートセンサの検知対象ガス濃度

スマートセンサの検知対象ガス濃度

表示例

注意 センサによっては、検知対象ガス以外のガスに対してもプラス方向に感度を有する場合があります。
それらのガスが共存する環境で使用した場合、実際に存在する検知対象ガス濃度よりも高く指示する
恐れがありますので、注意してください。例えば、シアン化水素（HCN）用の定電位電解式センサでは、
硫化水素（H2S）に対して HCN の約 2 倍の感度を有しますので、H2S が共存する環境下では特に注意が必要です。
センサがプラス方向に感度を有して干渉するガス例
使用するセンサの検知原理（検知対象ガス）／

干渉ガス

定電位電解式（HCN）
定電位電解式（HCN）
定電位電解式（HCN）
定電位電解式（SO2）定電位電解式（SO2）定電位電解式（CO）
定電位電解式（Cl2）
定電位電解式（Cl2）
定電位電解式（PH3）定電位電解式（PH3）定電位電解式（PH3）
接触燃焼式（HC/CH4）
赤外線式（HC/CH4）
PID 式（VOC）

／
／
／
／
／
／
／
／
／
／
／
／
／
／

H2S
SO2
C2H2
H2
CO
H2
SO2
HCｌ
SO2
HCN
H2S
全ての可燃性ガス
炭化水素系の可燃性ガス
全ての VOC- 48 -

5 使用方法

5-6. 検知する注意 定電位電解式センサは、原理上干渉ガスの種類によって、マイナス方向に感度を有する場合があります。それらのガスが共存する環境では、実際に存在する検知対象ガス濃度よりも低く指示する恐れがありますので、注意してください。
定電位電解式センサがマイナス方向に感度を有して干渉するガス例
使用するセンサの検知原理（検知対象ガス）／

干渉ガス

定電位電解式（H2S）
定電位電解式（HCN）
定電位電解式（NO2）
定電位電解式（SO2）定電位電解式（NH3）
定電位電解式（PH3）

／
／
／
／
／
／

NO2
NO2
SO2
NO2
H2S
NO2

 シリコーン化合物、ハロゲン化物、硫黄化物などが存在する環境にてニューセラミック式センサを使用した場合、センサの寿命が短くなったり、センサの可燃性ガスに対する感度が低下し、正確な指示を得られなくなる恐れがありますので、注意してください。やむを得ず使用する場合は使用時間を極力短くし、使用後は新鮮な空気を吸引し、指示に異常がないことを確認してください。
 ハロゲン化炭化水素や硫化物が高濃度、または低濃度でも長時間存在する環境にてガルバニ電池式センサを使用した場合、センサの寿命が短くなったり、センサの酸素に対する感度が低下し、正確な指示を得られなくなる恐れがありますので、注意してください。やむを得ず使用する場合は使用時間を極力短くし、使用後は新鮮な空気を吸引し、指示に異常がないことを確認してください。
 本器のニューセラミック式可燃性ガスセンサ＜%LEL＞が正確なガス検知および濃度表示をするには、
ある一定以上の酸素濃度が必要です。
 イナートガス中の酸素濃度を長時間測定する場合は、二酸化炭素濃度は必ず 15%以下の雰囲気で使用してください。15%より高い二酸化炭素濃度の雰囲気で使用する場合は、極力測定する時間を短くしてください。高濃度下で長時間使用すると、酸素センサの寿命が短くなる原因となります。
 高濃度可燃性ガスの検知を長時間続けると、可燃性ガスセンサ＜%LEL＞に悪影響を及ぼすことがあります。あらかじめ高濃度の可燃性ガスが存在することが分かっている場所を測定する場合は、可燃性センサ＜%LEL＞保護設定（P.71）を ON にしてから使用してください。
 吸着性の高いガスを吸引してしまった場合には、新鮮な空気を吸引するなどガス採集チューブのクリー
ニングを行い、指示値がゼロに戻ったことを確認してから使用してください。
 特定の毒性ガスや溶剤系 VOC など、吸着性の高いガスを測定する際にオプションのガス採集チューブ（0.75m/5m/10m/20m/30m）を使用した場合、チューブ内にガスが吸着し、測定ポイントに実際に存在する検知対象ガス濃度よりも低く指示する恐れがありますので、注意してください。
 高濃度の VOC ガスと接触した場合、CO センサの指示値が上昇する可能性があります。指示値が上昇して戻らない場合、CO センサ用の活性炭フィルタの交換が必要です。活性炭フィルタの交換については
弊社営業部までご連絡ください。
 塩素（Cl2）およびアンモニア（NH3）において、使用温度範囲の下限温度（-20℃）付近では、ガスの性状により応答が遅くなる恐れがありますので注意してください。
 PID 式 VOC センサは、高濃度のメタンガス、エタンガス、プロパンガスなどを吸引した場合、濃度表示部に「－－－－」を表示し、ランプが点滅してブザーが鳴り、一時的に測定できなくなることがあります。これ
らのガスが存在する環境では、濃度表示部に「－－－－」が表示されない場合でも、VOC 濃度を正しく測定できない恐れがありますので注意してください。なお、VOC センサの濃度表示部に「－－－－」が表示されている場合でも、VOC センサ以外の影響を受けないセンサは継続して測定できます。
PID 式 VOC センサの濃度表示部に「－－－－」が表示される干渉ガス例
干渉ガス

／

ガス濃度

メタンガス
エタンガス
プロパンガス／／／

6vol%以上
80vol%以上
90vol%以上- 49 -

5 使用方法

5-6. 検知する

注記 低温度の環境下では、電池の性能上、使用時間が短くなります。
 低温時は、LCD 表示部の応答が遅くなる場合があります。
 100%LEL 以上の高い濃度の可燃性ガスを吸引した場合は、テーパーノズルやフィルタ内に吸着したガスが残っている恐れがあります。高濃度の可燃性ガスを吸引した後には必ず新鮮な空気を吸引して、指示値がゼロ付近になるまでエアクリーニングを行い、吸着ガスを除去してください。完全にクリーニングされる前にゼロ校正を行うと、正確な調整とならず、測定に影響を及ぼす可能性があります。
 赤外線式可燃性ガス（HC および CH4）センサでは、検知した可燃性ガス濃度が 100%LEL を超えると、自動的に vol%レンジに切り替わります。また濃度が低下して来ると再び%LEL レンジに復帰します。下記に切替タイミングの例を示します
ガス濃度とレンジ切替タイミングチャート例
HC/CH4 濃度

100%LEL(HC:1.8vol%)(CH4:5.0vol%)%LEL レンジ

vol%レンジ

切替ポイント%LEL レンジ

切替ポイント※可燃性ガス濃度表示について
HC：イソブタン換算表示
CH4：メタン表示
レンジ切替ポイントは、ガスの爆発下限界値です。イソブタンの場合は 1.8vol%、メタンの場合は 5.0vol%です。

- 50 -

5 使用方法

5-6. 検知する

5-6-3．ベンゼンセレクトモード（VOC<10.0eV>センサを搭載している仕様のみ）
ベンゼンセレクトモードを起動し、プレフィルターチューブ及びチューブホルダーを装着した状態でテーパーノズルを検知する場所に近づけ、LCD 表示部の測定結果を読み取ってください。
BNZ

TUBE

5.20
再測定︖

ベンゼン濃度ppm

進む︓ENTER
戻る︓DISP
STEL︓RESET

表示例

注意



測定時間は温度によって異なります。一定の測定時間経過後の測定結果を読み取ってください。測定
方法については「ベンゼンセレクトモードへの切替」（P.59）を参照してください。
ベンゼンセレクトモード中は、VOC センサ以外のセンサは作動せず、ガス警報も発報しません。

- 51 -

5 使用方法

5-6. 検知する

5-6-4. スナップロガ
測定中の任意の瞬時値を最大 256 点まで記録することができます。
データ記録数が最大になった場合は、最も古いデータから上書きしていきます。

1

ノーマルモードの測定画面で、▲ ボタンと ▼ ボ
タンを同時に長押しする
以下のように、メモリ番号、ステーション ID、記録日時、
記録濃度が繰り返し表示されます。

2

ENTER ボタンを押す
画面に「セーブ」と表示され、ENTER ボタンを押したときのメモリ番号、ステーション ID、日時、ガス濃度が記録されます。
記録後、再びメモリ番号～記録濃度が繰り返し表示されます。続けて記録する場合は、ENTER ボタンを押して
ください。

3

終了する場合は、DISP ボタンを押す
測定画面に戻ります。

注記
 スナップロガで記録したガス濃度のデータは、「スナップロガ表示」（P.68）で確認できます。

- 52 -

5 使用方法

5-7. 電源を切る

5-7. 電源を切る
注意


測定終了後、濃度表示がゼロ（酸素濃度表示の場合、20.9%）に戻っていない場合は、新鮮な大気中に放置して表示がゼロに戻ってから電源を切ってください。

POWER/ENTER ボタンを押し続ける
電源を切るときは、安全な場所で、表示がゼロ（0、酸素は20.9％）に戻ってから、POWER/ENTER ボタンを長押しします。

ブザーが “ピッピッピッ” と 3 回鳴り、表示部に「電源オフ」と表示され、電源が切れます。

電源オフ表示注記 電源を切る際は、表示が消えるまで押し続けてください。

注意

機器が汚れている場合は、ウエスなどで拭き取ってください。
機器の汚れを拭き取る際は、アルコールやベンジンなどの有機溶剤を使用しないでください。

- 53 -

6 設定方法

6-1. DISP モードのフロー

6
6

設定方法

6-1. DISP モードのフロー
DISP モードでは、各種表示の確認と変更などを行うことができます。

ノーマルモードの測定画面で、DISP ボタンを押す
DISP ボタンを押すたびに、各種画面が順番に表示されます。

各種設定終了後は、DISP ボタンを押す
前の画面に戻ります。

ノーマルモードの測定画面に戻る場合は、RESET ボタンを押す
各種画面において RESET ボタンを押すと、測定画面に戻ります。

項目内容

ベンゼンセレクトモード
への切替

専用のプレフィルターチューブ（CF-8338）を使用し、ベンゼンを測定するベンゼンセレクト
モードに切り替えます。

（VOC＜10.0eV＞センサを搭載している仕様のみ表示）

PID ガス選択設定（VOC<10.0eV>センサ）（VOC＜10.0eV＞センサを搭載している仕様のみ
表示）

本器にあらかじめ登録してあるガスに設定変更することで、VOC＜10.0eV＞センサの検知
対象ガス（イソブチレン）から読み替えた濃度を表示します。

LCD 表示備考

TUBE

ENTER ボタンを押して、ベンゼ
ン測定画面へ
（P.59）

ﾍﾞﾝｾﾞﾝ測定

進む︓ENTER
戻る︓DISP
測定︓RESETPID 設定ｶﾞｽ
Isobutylene
10.0eV
変更︓ENTER
保持︓DISP
測定︓RESET- 54 -ENTER ボタンを押して、設定画
面へ（P.61）

6 設定方法
PID ガス選択設定（VOC<10.6eV>センサ）（VOC＜10.6eV / ppb＞,VOC＜10.6eV / ppm＞センサを搭載している仕様のみ表示）6-1. DISP モードのフロー
本器にあらかじめ登録してあるガスに設定変更することで、VOC＜10.6eV / ppb＞センサ、VOC＜10.6eV / ppm＞センサの検知対象ガス（イソブチレン）から 読み 替 え た濃 度 を 表 示します。

PEAK 値表示／クリア

電源を入れてから現在までの間で検知された、ガスの最高濃度値（酸素の場合は最低濃度）を表示します。

STEL 値表示

電源を入れてから現在までのSTEL 値を表示します。

TWA 値表示

電源を入れてから現在までのTWA 値を表示します。

フルスケール／警報点表
示／警報テスト

フルスケール値および各種警報設定値の表示、また表示した
設定における警報動作を確認できます。

- 55 -PID 設定ｶﾞｽ
Isobutylene

ENTER ボタンを押して、設定画
面へ（P.61）

10.6eV
変更︓ENTER
保持︓DISP
測定︓RESET
表示／クリア画
面へ（P.63）

進む︓ENTER
戻る︓DISP
測定︓RESETENTER ボタンを押して、確認画面へ（P.64）

6 設定方法

6-1. DISP モードのフロー

測定経過時間表示

電源を入れてからの測定時間を表示します。

日時と電圧表示

日付と時刻、電池残量と電池種別を表示します。

データロガ残り時間表示

データロガで記録できる残り時間を表示します。

データロガ消去

スナップロガで記録したデータをクリアします。
進む︓ENTER
戻る︓DISP
測定︓RESET

ユーザーID 表示／選択

ユーザーID を表示および選択します。

- 56 -ENTER ボタンを押して、クリア
画面へ（P.65）ENTER ボタンを押して、表示および選択画面へ（P.66）

6 設定方法

6-1. DISP モードのフロー

ステーション ID 表示／
選択

ステーション ID を表示および選択します。

ENTER ボタンを押して、表示および選択画面へ（P.67）

スナップロガ表示

スナップロガで記録したデータを表示します。

ENTER ボタンを押して、表示画面へ（P.68）

進む︓ENTER
戻る︓DISP
測定︓RESET

ピークバー表示設定

測定ガススクロール表示
設定

表示上下反転設定

測定画面のガス濃度値右側に表示されるバー表示において、ピーク値を点滅させる設定です。

測定画面の 6 分割表示または 1
分割表示か設定できます。
また、1 分割表示の場合、表示の切り替えを自動または手動か設定できます。

進む︓ENTER
戻る︓DISP
測定︓RESET

進む︓ENTER
戻る︓DISP
測定︓RESET

本器の向きにより、LCD 表示の上下を 180 度反転します。

進む︓ENTER
戻る︓DISP
測定︓RESET- 57 -ENTER ボタンを押して、設定画
面へ（P.69）ENTER ボタンを押して、設定画
面へ（P. 70）

ENTER ボタンを押して、設定画
面へ（P.71）

6 設定方法
可燃性センサ＜%LEL＞
保護設定

6-1. DISP モードのフロー
高濃度の可燃性ガスから可燃性センサ＜%LEL＞を保護します。

ENTER ボタンを押して、設定画
面へ（P.72）LCD 表示の白黒表示を反転します。

ENTER ボタンを押して、設定画
面へ（P.73）（可燃性ガス＜%LEL＞
を検知対象とする仕様
のみ表示）

表示白黒反転設定

進む︓ENTER
戻る︓DISP
測定︓RESET

言語設定

他の言語表示に設定している
場合に、英語表示に戻します。

（英語以外の言語を選択している場合のみ表示）

LANGUAGE
CHANGETO ENGLISH
YES:ENTER
NO:DISPEXIT:RESET

注記
 何も操作されない状態が約 20 秒間続くと、測定画面に戻ります。
 各種画面表示中に RESET ボタンを押すと、測定画面に戻ります。
 「言語設定」の画面で DISP ボタンを押すと、測定画面に戻ります。

- 58 -ENTER ボタンを押して、設定画
面へ（P.74）

6 設定方法

6-2. 各種設定

6-2. 各種設定
ベンゼンセレクトモードへの切替（VOC＜10.0eV＞センサを搭載している仕様のみ）
ベンゼンセレクトモードに切り替えます。ベンゼンセレクトモードでは、専用のプレフィルターチューブ（CF-8338）及びチューブホルダー（GF-284）を使用します。装着方法については「プレフィルターチューブ（CF-8338）及び
チューブホルダー（GF-284）の装着」（P.33）を参照してください。

1

DISP ボタンを押して右画面を表示

TUBE

ﾍﾞﾝｾﾞﾝ測定

させ、ENTER ボタンを押す

進む︓ENTER
戻る︓DISP
測定︓RESET

2

プレフィルターチューブ（CF-8338） 及びチューブホルダー（GF-284）を GX6000 に装着し、 ENTER ボタンを押す

TUBE

ﾍﾞﾝｾﾞﾝ測定

ﾁｭｰﾌﾞとﾎﾙﾀﾞをｾｯﾄして下さい
進む︓ENTER
戻る︓DISP

ポンプ及び履歴/トレンド機能が停止します。

3

ENTER ボタンを押す

TUBE

ベンゼン測定を行う場合は ENTER ボタンを押します。
測定を行わない場合は DISP ボタンを押し、ホルダを外して ENTER ボタンを押してください。ノーマルモードの測定画面に戻ります。

ポンプが作動し、ベンゼン測定を開始します。
測定時間がカウントダウン表示されます。
測定時間は温度によって異なります。
測定時間については下記一覧を参照してください。なお、下記番号は測定画面左下に表示されます。
1.
2.
3.
4.
5.
6.

-20.0
-10.0
0.0
+10.0
+20.0
+30.0

～ -10.1℃～-0.1℃～ +9.9℃～ +19.9℃～ +29.9℃～ +50.0℃

TUBE

ﾍﾞﾝｾﾞﾝ測定

BNZ

ﾍﾞﾝｾﾞﾝ測定

測定開始

ﾎﾙﾀﾞを外す

進む︓ENTER
戻る︓DISP

進む︓ENTER
戻る︓DISP

TUBE

4.80
測定中

BNZ

ppm

5

- 59 -

45 秒

4.80
ppm

ｷｬﾝｾﾙ︖
進む︓ENTER
戻る︓RESET中⽌︓RESET：135 秒：110 秒
：90 秒
：70 秒
：45 秒
：35 秒

TUBE

5

44 秒

6 設定方法

6-2. 各種設定

カウントダウン終了後、測定結果を表示します。
・再度ベンゼン濃度を測定する場合：
ENTER ボタンを押す
プ レ フ ィ ル ター チ ュ ー ブ を 交 換 し 、
ENTER
ボタンを押してください。
⇒手順 3 「測定開始」画面へ
・ノーマル測定に戻る場合：
DISP ボタンを押す⇒手順 3 「ホルダを外す」画面へ
・ベンゼンの STEL を測定する場合：
RESET ボタンを押す⇒手順 4 「STEL 測定開始」画面へ
4

BNZ

TUBE

TUBE

5.20
再測定︖

ppm

進む︓ENTER
戻る︓DISP
STEL︓RESETENTER ボタンを押す

ﾍﾞﾝｾﾞﾝ測定
ﾁｭｰﾌﾞ交換
進む︓ENTER
戻る︓DISP
測定︓ENTER

TUBE

ﾍﾞﾝｾﾞﾝ測定

STEL 測定を開始します。

STEL
測定開始
進む︓ENTER
戻る︓DISP

画面下に測定時間をカウントダウン表示します。測定時間は 900 秒です。

TUBE
BNZ STEL

0.00

測定を中断する場合は、 RESET ボタンを押し、そして ENTER ボタンを押します。

ppm

5.20
ppm

STEL 測定中
中⽌︓RESET
889 秒

測定結果を表示します。
・再度 STEL 測定を行う場合：
ENTER ボタンを押す⇒手順 3 「チューブ交換」画面へ
・ノーマル測定に戻る場合：
DISP ボタンを押す⇒手順 3 「ホルダを外す」画面へ

TUBE
BNZ STEL

3.20

再測定︖

ppm

進む︓ENTER
戻る︓DISP- 60 -TUBE
BNZ STEL

0.00
ppm

5.20
ppm

ｷｬﾝｾﾙ︖
中⽌︓ENTER
戻る︓DISP

6 設定方法

6-2. 各種設定

PID ガス選択設定（VOC センサを搭載している仕様のみ）
通常、揮発性有機化合物（VOC）の濃度表示はイソブチレン換算表示ですが、あらかじめ登録してあるガスに読
み替え、濃度表示することができます。

1

2

DISP ボタンを押して右画
面を表示さ
せ 、 ENTER
ボタンを押すPID 設定ｶﾞｽ
Isobutylene

PID 設定ｶﾞｽ
Isobutylene

10.0eV

10.6eV

変更︓ENTER
保持︓DISP
測定︓RESET

変更︓ENTER
保持︓DISP
測定︓RESETVOC＜10.0eV＞センサを搭載している仕様の場合

VOC＜10.6eV＞センサを搭載している仕様の場合▲ または ▼
ボタンを押して選択する
ユーザーリストは設定したガス
リスト、記録は最近選択したガスリストです。
A～X では全てのガスを表示します。

3

ENTER ボ
タンを押す
ガス種の名称が表示されます。
手順 2 に戻るには、 DISP ボタ
ンを押します。

4

ENTER ボ
タンを押す
各ガスの名称、
化学式、換算係数値などが表示されます。
手順 3 に戻るには、 DISP ボタ
ンを押します。

- 61 -

6 設定方法

56-2. 各種設定

ENTER ボタンを押す
設定終了後、
自動で手順 1の画面に戻りま
す。

注記







VOC＜10.0eV＞センサと VOC＜10.6eV＞センサ両方を搭載している仕様の場合は、両方のセンサの設定が必要です。VOC＜10.0eV＞センサ、VOC＜10.6eV＞センサの順で設定画面が表示されますので設定を行ってください。
電源を切っても設定は保持されます。
ユーザーリストには、よく選択するガス種を最大 30 種類まで登録できます。
ユーザーリストを利用するには、PID リスト設定プログラム（別売）が必要です。
記録には、全ガスリストから選択したガス種の履歴（最大 8 種類）を残すことができます。
読み替えできるガスの種類については付録の「揮発性有機化合物（VOC）読み替えリスト」（P.100）を参照
してください。

- 62 -

6 設定方法

6-2. 各種設定

PEAK 値表示／クリア
電源を入れてから現在までの間で、最高濃度の測定値（酸素では最低濃度）の表示およびクリアができます。

1

DISP ボタンを押して右画面を表示させる

2

ピーク値をクリアするときは、 RESET
ボタンを長押しする

3「RELEASE」が表示されたら、
RESET ボタンを離す

PEAK 値がクリアされました。
PEAK 値クリア後、自動で手順 1 の画面に戻ります。

- 63 -

6 設定方法

6-2. 各種設定

フルスケール/警報点表示/警報テスト
フルスケール値および各種警報設定値の表示、また表示した設定における警報動作を確認できます。
なお、警報テストでは LCD 表示は切り替わりません。

1

DISP ボタンを押して右画面を表示させ、ENTER ボタンを押す
進む︓ENTER
戻る︓DISP
測定︓RESET

2▲ または ▼ ボタン押し、フルスケール値
または各種警報設定値を表示させる

フルスケール表示

アラーム１表示

アラーム２表示

TWA 値表示3

任意の画面を表示し、 ENTER ボタンを押す
警報窓の赤ランプが点滅して、表示されている画面での警報動作を確認できます。

4

警報動作を止めるときは、ENTER ボタ
ンを押す
表示および警報テストを終了する場合、
DISP ボタンを押すと手順 1 の画面に戻ります。

- 64 -STEL 値表示

6 設定方法

6-2. 各種設定

データロガ消去
スナップロガで記録したログデータをクリアします。

1

DISP ボタンを押して右画面を表示さ
せ、ENTER ボタンを押す
進む︓ENTER
戻る︓DISP
測定︓RESET

2

ログデータをクリアするときは、
ENTER ボタンを押す
ログデータをクリアしない場合、DISP ボタ
ンを押すと手順 1 の画面に戻ります。

3

ENTER

ENTER ボタンを押す

ログデータが消去されました。
消去後、自動で手順 1 の画面に戻ります。

- 65 -

6 設定方法

6-2. 各種設定

ユーザーID 表示／選択
ユーザーID を表示および選択できます。

1

DISP ボタンを押して右画面を表示さ
せ、ENTER ボタンを押す

2▲ または ▼ ボタン押して、ユーザー
ID を選択する
ユーザーID を表示および選択しない場合、
DISP ボタンを押すと手順 1 の画面に戻ります。

3

ENTER ボタンを押す
選択終了後、自動で手順 1 の画面に戻ります。

注記 初めて使用されるときには、ユーザーID は右記のように表示されます。
 指定されていない場合のユーザーID 番号は 001～128 で登録されています。
 ID の登録および変更には、データロガマネジメントプログラム（別売）が必要になります。お求めになる場合、販売店または弊社営業所までお問い合わせください。

- 66 -

6 設定方法

6-2. 各種設定

ステーション ID 表示／選択
ステーション ID を表示および選択できます。

1

DISP ボタンを押して右画面を表示さ
せ、ENTER ボタンを押す

2▲ または ▼ ボタン押して、ステーション ID を選択する
ステーション ID を表示および選択しない場
合、DISP ボタンを押すと手順 1 の画面に戻ります。

3

ENTER ボタンを押す
選択終了後、自動で手順 1 の画面に戻ります。

注記 初めて使用されるときには、ステーション ID は右記のように表示されます。
 指定されていない場合のステーション ID 番号は 001～128 で登録されています。
 ID の登録および変更には、データロガマネジメントプログラム（別売）が必要になります。お求めになる場合、販売店または弊社営業所までお問い合わせください。

- 67 -

6 設定方法

6-2. 各種設定

スナップロガ表示
スナップロガで記録したログデータを表示します。

1

DISP ボタンを押して右画面を表示さ
せ、ENTER ボタンを押す
進む︓ENTER
戻る︓DISP
測定︓RESET
RESET
測定︓

2

▲ または ▼ ボタン押して、記録デー
タを選択する
記録データは、年月日、時間とメモリ番号で表示されます。また、ステーション ID を設定している場合、メモリ番号の下に表示されます。
ログデータを表示しない場合、DISP ボタン
を押すと手順 1 の画面に戻ります。

3

ENTER ボタンを押す
選択した記録データが表示されます。
もう一度、ENTER ボタンを押すと、手順 2の画面に戻ります。
ログデータの表示を終了する場合、 DISP
ボタンを押すと手順 1 の画面に戻ります。

注記 ガス濃度を記録する操作については、「スナップロガ」（P.52）を参照
してください。
 ガス濃度のデータが記録されていないときは、右記のように表示されます。

- 68 -

6 設定方法

6-2. 各種設定

ピークバー表示設定
測定画面のガス濃度値右側に表示されるバー表示において、ピーク値を点滅させる設定です。

1

DISP ボタンを押して右画面を表示さ
せ、ENTER ボタンを押す
進む︓ENTER
戻る︓DISP
測定︓RESET

2▲ または ▼ ボタンを押して選択する
バー表示におけるピーク値点滅の有無を選択します。
設定を変更しない場合、DISP ボタンを押すと手順 1 の画面に戻ります。

3

ENTER ボタンを押す
設定終了後、自動で手順 1 の画面に戻ります。

注記
 ピークバーを表示する設定を選択すると、右記のようにバー
表示のピーク値が点滅します。

- 69 -

6 設定方法

6-2. 各種設定

測定ガススクロール表示設定
測定画面の表示を複数ガス同時表示とするか、1 ガス表示とするかを選択できます。1 ガス表示の場合、表示の切り替えを自動で行うか手動で行うか選べます。

1

DISP ボタンを押して右画面を表示さ
せ、ENTER ボタンを押す
進む︓ENTER
戻る︓DISP
測定︓RESET

2▲ または ▼ ボタン押して、表示形式
を選択する「全て表示」は分割表示です。
「スクロール自動」は複数ガスを 1 ガスず
つ、自動で切り替えて順次表示します。
「スクロール手動」は 1 ガス表示で、ENTER
ボタンを押して、手動で他のガス濃度表示に切り替えます。
設定を変更しない場合、 DISP ボタンを押すと手順 1 の画面に戻ります。

3

ENTER ボタンを押す
設定終了後、自動で手順 1 の画面に戻ります。

注記 6 ガス表示と 1 ガス表示の例を示します。
 ガス濃度表示の設定は、電源の入/切でリセットされます。

- 70 -

6 設定方法

6-2. 各種設定

表示上下反転設定
本器の向きにより、LCD 表示の上下を 180 度反転します。

1

DISP ボタンを押して右画面を表示さ
せ、ENTER ボタンを押す

進む︓ENTER
戻る︓DISP
測定︓RESET

2▲ または ▼ ボタンを押して選択する
表示上下反転設定を選択します。
設定を変更しない場合、DISP ボタンを押すと手順 1 の画面に戻ります。

3

ENTER ボタンを押す
設定終了後、自動で手順 1 の画面に戻ります。

注記 表示上下反転設定でオフ（表示向き固定）を選択している場
合、右図のように画面右 上に「LOCK」を表示します（点灯）。
 表示上下反転設定でオン（表示向き反転）を選択していても、使用時に DISP ボタンを長押しすると、表示の向きを固定することができます。固定している場合、右図のように画面右上に「LOCK」を表示します（点滅）。
 DISP ボタンを長押しして表示の向きを固定した場合、電源の入/切で設定がリセットされます。

- 71 -

6 設定方法

6-2. 各種設定

可燃性センサ＜%LEL＞保護設定（可燃性ガス＜%LEL＞を検知対象とする仕様のみ）
可燃性センサ＜%LEL＞が高濃度可燃性ガスに接触してもダメージを受けないように、可燃性センサ＜%LEL＞の電源を切ります。

1

DISP ボタンを押して右画面を表示さ
せ、ENTER ボタンを押す

2▲ または ▼ ボタンを押して選択す
る
可燃性センサ＜%LEL＞保護の設定を選
択します。

3

ENTER ボタンを押す
設定終了後、自動で手順 1 の画面に戻ります。

注記
 ON 設定にしている場合、可燃性ガス＜%LEL＞濃度表示部に「－－－－」を表示します。また、時計表示部に「NO ALARM」と表示し、全てのガスにおいてガス警報機能が無効となります。

- 72 -

6 設定方法

6-2. 各種設定

表示白黒反転設定
LCD 表示の白黒表示を反転します。

1

DISP ボタンを押して右画面を表示さ
せ、ENTER ボタンを押す
進む︓ENTER
戻る︓DISP
測定︓RESET

2▲ または ▼ ボタンを押して選択する
表示白黒反転設定を選択します。
設定を変更しない場合は、DISP ボタンで
手順 1 の画面に戻ります。

3

ENTER ボタンを押す
設定終了後、自動で手順 1 の画面に戻ります。

注記
 白黒反転表示の例を示します。

- 73 -

6 設定方法

6-2. 各種設定

言語設定
他の言語表示で使用している場合に、英語表示に戻します。
誤った言語に変更してしまった場合などは、この機能を利用して英語表示に戻してから再設定してください。

1

DISP ボタンを押して右画面を表示さ
せ、ENTER ボタンを押す

CHANGETO ENGLISH
YES:ENTER
NO:DISPEXIT:RESET

2

ENTER ボタンを押す
英語表示にしない場合、 DISP ボタンを押すと手順 1 の画面に戻ります。
CHANGETO ENGLISHCONFIRM ?

言語表示が英語に切り替わります。
設定後、自動で手順 1 の画面（英語表示）を表示します。

注記 英語表示から日本語表示に戻すには、ユーザーモード（P.75）から切り替えてください。DISP モードから日本語表示に戻すことはできません。

- 74 -

6 設定方法

6-3. ユーザーモード設定

6-3. ユーザーモード設定
ユーザーモードでは、日時表示やガス濃度の表示位置などを、使いやすいように変更できます。

ユーザーモードを表示する
1

電源を切った状態で、▲ および ▼ ボタンを押しながら、POWER ボタンを押す
ユーザーモードとメンテナンスモードの選択画面が表示されます。

2「ユーザーモード」を選択している画面で、ENTER
ボタンを押すユーザーモードのメニューが表示されます。

3

設定終了後は、ユーザーモードメニューで「測定開始」を選択して ENTER ボタンを押す
電源を入れたときと同様に作動して、測定画面に進みます。

注記 各種設定後はユーザーモードメニューに戻ります。設定の途中で戻るときは、DISP ボタンを押してください。
 メンテナンスモードは、正常に測定を行うための重要な設定を行うモードです。誤って設定を変更してしまう
ことを防止するため、お客様には操作していただけません。万一、メンテナンスモードを選択してしまった場
合には、一度電源を切り、再度電源を入れ直してください。

- 75 -

6 設定方法

6-3. ユーザーモード設定

日付／時刻を設定する
日付と時刻を設定します。

1▲ または ▼ ボタンを押して「日時」を選択する

2

ENTER ボタンを押す「年」の部分が点滅します。
▲ または ▼ ボタンで数字を変更できます。

3

年を設定したら ENTER ボタンを押す「月」の部分が点滅します。
▲ または ▼ ボタンで数字を変更できます。
同様にして、日、時、分を設定します。
DISP ボタンを押すと月→年のように前に
戻れます。

4

分を設定し、ENTER ボタンを押す
設定終了後、自動でユーザーモードメニューに戻ります。

- 76 -

6 設定方法

6-3. ユーザーモード設定

日付の表示フォーマットを選択する
日付の表示を 3 種類のフォーマットから選択できます。

1▲ または ▼ ボ タン を押 し て「日 時フォーマット」を選択し、ENTER ボタンを押す

2▲ または ▼ ボタンを押して表示を選択する「年/月/日」、「月/日/年」、「日/月/年」から表示フォーマットを選択します。
表示フォーマットを変更しない場合、 DISP
ボタンで手順 1 の画面に戻ります。

3

ENTER ボタンを押す
設定終了後、自動でユーザーモードメニューに戻ります。

- 77 -

6 設定方法

6-3. ユーザーモード設定

ガス濃度値の表示位置を入れ替える
測定ガス濃度値の表示位置を入れ替えることができます。

1▲ または ▼ ボタンを押して「チャネル表示切替」を選択し、ENTER
ボタンを押す2▲ または ▼ ボタンを押して変更する
表示位置を選択し、 ENTER ボタンを押す
CH1～CH6 の表示位置は以下になります。

「戻る」でユーザーモードメニューに戻ります。

3▲ または ▼ ボタンを押して入れ替え
る表示を選択する
選択した成分と、手順 2 で選択した成分（点滅）の位置が入れ替わります。

4

ENTER ボタンを押す
設定後、手順 2 の画面に自動で戻ります。
ユーザーモードメニューに戻る場合は、
DISP ボタンを押す、または「戻る」を選択
し、ENTER ボタンを押します。

注記
 複数の CH 位置に同じ測定ガス表示を割り付けることはできません。

- 78 -

6 設定方法

6-3. ユーザーモード設定

表示言語を変更する
LCD 表示の表示言語を変更します。

1▲ または ▼ ボタンを押して「言語」を選択し、ENTER ボタンを押す

2▲ または ▼ ボタンを押して言語を選択する
ENGLISH＞ JAPANESEITALIANSPANISHGERMANFRENCHPORTUGUESE
RUSSIANKOREAN

3

ENTER ボタンを押す
設定終了後、選択した言語に表示が替わ
り、自動でユーザーモードメニューに戻ります。

- 79 -

6 設定方法

6-3. ユーザーモード設定

ROM/SUM を表示する
ROM 番号と、データと一緒に送信する誤り検出用データ（チェックサム）のバージョンを確認できます。

1▲ または ▼ ボタンを押して「ロム・サム」を選択し、ENTER ボタン
を押す

ROM 番号が表示されます。

計算後、SUM が表示されます。

2

ENTER ボタンを押す
表示終了後、自動でユーザーモードメニューに戻ります。

- 80 -

6 設定方法

6-3. ユーザーモード設定

パスワードを設定する
ユーザーモードに入るためのパスワードを設定します。

1▲ または ▼ ボタンを押して「パスワード」を選択し、 ENTER ボタ
ンを押す

2▲ または ▼ ボタンを押して ON を選
択し、ENTER ボタンを押す

3

4 桁のパスワードを設定する
左端の「0」が点滅します。
▲ または ▼ ボタンを押して 0～9 を選択
し、ENTER ボタンを押すと、次の桁が点滅します。

4

ENTER ボタンを押す
設定終了後、自動でユーザーモードメニューに戻ります。

注記 パスワードを設定すると、ユーザーモードに入る際に、右記のようなパスワード入力画面が表示されます。▲ または ▼ ボタンでパス
ワードを入力して、ENTER ボタンを押してください。

- 81 -

7 保守点検7-1. 点検の頻度と点検項目

7
7

保守点検

本器は防災・保安上重要な計器です。
本器の性能を維持し、防災・保安上の信頼性を向上するために、定期的な保守・点検を実施してください。

7-1. 点検の頻度と点検項目ご使用になる前に、以下の項目を定期的に点検してください。
・日常点検：作業前に点検します。
・1 ヶ月点検：1 ヶ月に 1 回、警報テストを行って点検します。
・定期点検：保安機器としての性能を維持するため、6 ヶ月に 1 回以上の頻度で点検します。

点検項目

点検内容

日常
点検

1ヶ
月点
検

定期
点検

電池残量

電池残量が十分であるか確認してください。

○

○

○

濃度表示

新鮮な空気を吸引させて濃度表示値が 0（酸素計では 20.9%）であることを確認してください。0 でない場合は、周囲に雑ガスがない
ことを確認してゼロ校正を行ってください。

○

○

○

本体動作

LCD 表示を確認し、故障表示がないか確認してください。

○

○

○

ポンプ動作

ポンプの動作状態表示を確認し、故障表示がないか確認してくだ
さい。

○

○

○

フィルタ

フィルタに汚れがないか確認してください。

○

○

○

警報テスト

警報テストを実施し、警報窓のランプおよびブザー、バイブレー
ターが正常に動作するか確認してください。

－

○

○

ガス感度校正試験用標準ガスを用いて感度校正をしてください。

－

－

○

ガス警報確認試験用標準ガスを用いてガス警報の確認を行ってください。

－

－

○

警告


万一、本器に異常が見つかった場合は、速やかに販売店または最寄りの弊社営業所までご連絡ください。

注記 試験用標準ガスによるガス感度校正は、6 ヶ月に一度は行ってください。
 ガス感度校正を行うには専用の器具や試験用標準ガスの作製が必要になります。したがって、ガス感度校
正については、販売店または最寄りの弊社営業所までご連絡ください。
 本器に内蔵しているセンサは有効期限があり定期的に交換が必要です。
 ガス感度校正の際、校正できない、ゼロ校正しても指示が戻らない、指示がふらつく等の症状が出たら、センサの寿命です。販売店または最寄りの弊社営業所までご依頼ください。なお、保証期間は 1 年です。
- 82 -

7 保守点検7-1. 点検の頻度と点検項目

メンテナンスサービスについて
弊社では、ガス感度校正などを含めた定期点検、調整、整備などに関するサービスを行っております。
試験用標準ガスを作製するには、所定濃度のガスボンベや、ガス袋など専用器具が必要となります。
弊社指定のサービス員は、作業を行う上での専用器具や、その他製品に関する専門知識などを備えたスタッフで構成されております。機器の安全動作を維持するために、弊社メンテナンスサービスをご利用いただきますようお願いします。
メンテナンスサービスの主な内容を以下に記します。詳細は、弊社営業所までお問い合わせください。

＜主なメンテナンスサービスの内容＞
電池残量の確認

電池残量の確認を行います。

濃度表示の確認

ゼロガスを用いて濃度表示値がゼロ（酸素計では 20.9%）であることを確認します。
指示がずれている場合はゼロ校正（ゼロ調整）を行います。

流量の確認

外部の流量計を用いて、流量の確認を行います。

フィルタの確認

ダストフィルタの汚れ具合や目詰まりがないかを確認します。
汚れが目立つ場合や、目詰まりを起こしている場合は交換します。

警報テスト

警報テストを実施し、警報ランプおよびブザー、バイブレーターが正常に動作することを確認します。

ガス感度校正試験用標準ガスを用いて感度校正を行います。

ガス警報確認試験用標準ガスを用いてガス警報を確認します。
 警報確認（警報設定値に達した際に警報の発信を確認）
 遅れ時間確認（警報を発信するまでの遅れ時間を確認）
 ブザー、ランプ、バイブレーター、濃度表示の確認（警報 2 段階、それぞれの動作を確認）

機器の清掃・修繕（目視診断）

機器外観の汚れや傷を確認し、目立った箇所を清掃・修繕します。
亀裂や破損がある場合は、部品を交換します。

機器の操作確認ボタン操作をして各種機能の動作確認や、パラメーターなどをチェックします。

劣化部品の交換

センサやフィルタ、ポンプなど、劣化部品を交換します。

- 83 -

7 保守点検

7-2. 清掃方法

7-2. 清掃方法
本器が著しく汚れた場合は清掃を行ってください。清掃する際は、必ず電源を切った状態で、ウエスなどで汚れを拭き取ってください。水拭きや有機溶剤を使用して清掃すると、故障の原因となりますので使用しないでください。
テーパーノズルが著しく汚れている場合は、ガス検知に影響を及ぼす可能性があるので、ドライエアなどでクリー
ニングを行ってください。

注意

本器の汚れを拭き取る際は、水をかけたり、アルコールやベンジンなどの有機溶剤を使用しないでください。本器の表面の変色や損傷およびセンサの故障の原因となります。

注記
 本器が濡れた後は、ブザー放音口や溝に水が溜まっている場合があります。以下の手順で水抜きを行ってください。
① 本器に付着した水分を乾いたタオル、布などでよく拭き取る② 本器をしっかり持ち、ブザー放音口を下に向けて 10 回程度振る③ 内部から出てきた水分をタオル、布などでよく拭き取る④ 乾いたタオル、布などを下に敷き、常温で放置する- 84 -

7 保守点検

7-3. 各部品の交換

7-3. 各部品の交換
7-3-1. ガス吸引部のフィルタ交換
ガス吸引部にはダストフィルタ、金網フィルタが組み込まれています。フィルタは、使用している間に、汚れたり、詰まったりすることがあり、使用状況に応じて交換する必要があります。特に、ダストフィルタは水を吸った場合や流量が下がった場合、汚れが目立ってきた場合には必ず交換してください。交換用のフィルタについては、定期交換
部品（P.84）を参照してください。

1
2
3

フィルタケースを反時計方向に回して外す
フィルタを取り出し、新しいフィルタに交換する
外したフィルタケースを取り付ける

注記 ダストフィルタと金網フィルタは本体側に組み込まれています。
 弊社指定のフィルタ以外、使用しないでください。

- 85 -

7 保守点検

7-3. 各部品の交換

7-3-2. 定期交換部品
本器の消耗品は、下記の通りです。推奨周期を目安に消耗品を交換してください。

＜推奨交換部品リスト＞
推奨点検周期

推奨交換周期

数量（個/台）

活性炭フィルタ

3 ヶ月

6 ヶ月

1

CO センサ用。※

硫化水素除去フィルタ

3 ヶ月

6 ヶ月

1

可燃性センサ（%LEL 用）。※調湿フィルタ

3 ヶ月

6 ヶ月

1

H2S センサ用。※

活性炭フィルタ（CF-8350）

6 ヶ月

1年

1

VOC センサ用。
VOC センサ搭載時に使用。

活性炭フィルタ（CF-8501）

6 ヶ月

1年

1

VOC センサ用。
VOC センサ及び CO2 センサ
搭載時に使用。

CO2 除去フィルタ（CF-284）

6 ヶ月

1年

1

CO2 センサ用。

ダストフィルタ

3 ヶ月

6 ヶ月

1

10 枚セットでの販売となります。

-

-

1

10 枚セットでの販売となります。

6 ヶ月

1～2 年

1

※

パッキン類

-

2年

1式

※

リチウムイオン電池ユニット（BUL-6000）-

充放電 500
回

1

リチウムイオン電池ユニットをご使用の場合。

アルカリ乾電池--

3

アルカリ乾電池ユニットをご使用の場合。単 3 形。

名称

金網
ポンプユニット（RP-12）

備考※ 部品交換後に専門のサービス員による動作確認が必要です。機器の安定動作と安全上、専門のサービス員にお任せ願います。販売店または最寄りの弊社営業所にご依頼ください。

注記
 上記の交換周期は目安であり、使用条件によって異なる場合があります。また、保証期間を表すものではありません。交換時期は定期点検の結果により変動することがあります。

- 86 -

8 保管および廃棄について

8-1. 保管または長期使用しない場合の処置

88保管および廃棄について

8-1. 保管または長期使用しない場合の処置
本器は下記の環境条件内で保管してください。
・常温、常湿、直射日光の当たらない暗所
・ガス、溶剤、蒸気などの発生しない場所
本器が収納されている梱包箱がある場合は、それに入れて保管してください。
梱包箱がない場合は、埃などを避けて保管してください。

注意




本器を長期間使用しない場合は、リチウムイオン電池ユニットは、本体から外して保管してください。また乾電池ユニットで使用されている場合は、乾電池を外してください。乾電池の液漏れにより、火災、ケガなどの原因となることがあります。
本器を長期間使用しない場合でも、6 ヶ月に一度は電源を入れ、ポンプが吸引することを確認してください（3 分間程度）。長期間動作させずにいると、ポンプのモータ内のグリスが固まり動作しなくなることが
あります。

注記
 リチウムイオン電池ユニットを装着したままで長時間使用しない場合は、電池マークが一つになる程度まで放電して保管することをお勧めします。満充電のまま保管すると、電池寿命が短くなるなど、電池の劣化が早まる可能性があります。
 乾電池ユニットで使用している場合、短期間使用しないときは、乾電池を入れたままで保管してください。本器は電源を切っているときでもセンサには常時通電されていますので、乾電池を入れたままでの保管が必要となります。

8-2. 再度使用する場合の処置
本器を保管後、再度使用する場合は、ガス校正を行ってください。

注意

ガス校正を含め、再調整は、販売店または最寄りの弊社営業所までご連絡ください。
保管場所と使用場所の温度が 15℃以上急変するような場合は、電源を入れた状態で使用場所と同様の環境下で 10 分程度馴染ませ、新鮮な大気中でゼロ校正を行ってから使用してください。

- 87 -

8 保管および廃棄について

8-3. 製品の廃棄

8-3. 製品の廃棄
本器を廃棄する場合は、産業廃棄物（不燃物）として地域の法令などに従い、適切な処理をしてください。

警告定電位電解式センサ、ガルバニ電池式センサには電解液が入っていますので、絶対に分解しないでくだ
さい。電解液に触れると皮膚がただれる恐れがあり、眼に入ると失明する恐れがあります。また、衣服に付着した場合、変色したり、穴が開く恐れがあります。
万一、電解液に触れた場合は、触れた部分を直ちに水で十分洗浄してください。乾電池を廃棄する場合
は、地域ごとに定められた方法に従って処分してください。

＜ＥＵ加盟各国内での廃棄について＞
EU 加盟各国内で、本器を廃棄する際は電池を分別してください。
リチウムイオン電池ユニット（BUL-6000）から取り外した電池、乾電池ユニット（BUD-6000）で使用した乾電池については、EU 加盟各国内の法令などに従い、各地域の分別収集システムやリサイクル制度に従い、適切な処理をしてください。

注記
クロスドアウトリサイクルダストビンマークについて
 このシンボルマークは、EU 電池指令 2006/66/EC に該当する電池を内蔵している製品に表示されており、電池を適切な方法で廃棄していただく必要があります。このシンボルマークは、電池を廃棄する際に一般ゴミとは分別して処理する必要があることを意味しています。

- 88 -

9 トラブルシューティング

9-1. 機器の異常

99

トラブルシューティング

このトラブルシューティングは、全ての不具合の原因を記載したものではありません。よく発生する不具合の原因究明の手助けとなるものを簡単に記載しています。
ここに記載されていない症状や対策を行っても復旧しない場合は、販売店または最寄りの弊社営業所までご連絡ください。

9-1. 機器の異常症 状＜画面表示＞

原因処

電池が極端に消耗している
電源が入らない置

リチウムイオン電池ユニットの場合：安全な場所で充電してください。
乾電池ユニットの場合：安全な場所で 3 本とも新品の乾電池に交換してください。

POWER ボタンを押す時間が不適切

電源を入れるときは POWER ボタンを押し、ピッと音が鳴ったら指を離してください。

電池ユニットの実装不良

電池ユニットが正しく本体に装着されているか確認
してください。

異常な動作をする

突発的な静電気ノイズなどによる影響

いったん電源を切り、再度電源を入れて（再起動）く
ださい。

操作ができない

突発的な静電気ノイズなどによる影響

安全な場所で、いったん電池ユニットを外してから、再度電池ユニットを取り付け、電源を入れて操作してください。

電池電圧低下警報が表示されている＜FAIL BATTERY＞

充電できない（リチウムイオン電池
ユニットのみ）電池残量がなくなっている

リチウムイオン電池ユニットの場合：電源を切り、安全な場所で充電を行ってください。
乾電池ユニットの場合：電源を切り、安全な場所で新品の乾電池に交換してください。

アダプタの接続が正しくない

AC アダプタの AC プラグおよび DC プラグを正しく差し込んでください。

充電回路に異常がある

販売店または最寄りの弊社営業所まで修理をご依頼ください。

満充電されている

満充電状態から再度充電を行っても、充電確認用
ランプは点灯しません。

- 89 -

9 トラブルシューティング症 状＜画面表示＞

流量低下警報が表示されている＜FAIL LOW FLOW＞

ゼロ校正ができない＜SENSOR FAIL＞

センサ異常＜SENSOR FAIL＞

システム異常＜FAIL SYSTEM＞

9-1. 機器の異常

原

因

処

置

水、油などを吸い込んでいる

テーパーノズルに損傷や水、油などの吸い込み跡がないか、確認してください。

フィルタが詰まっている

フィルタの取り付け状態、および詰まりやねじれなどを確認してください。

ポンプが劣化している販売店または最寄りの弊社営業所までポンプ交換をご依頼ください。

長期間動作させずに保管している（6 ヶ月以上）

流量低下警報が表示されたら、いったん電源を切り、再度電源を入れて（再起動）ください。数回この操作を行って改善しない場合は、販売店または最寄りの弊社営業所までポンプ交換をご依頼ください。

本器の周囲に新鮮な空気を供給していない

新鮮な空気を供給してください。

センサ感度の劣化販売店または最寄りの弊社営業所までセンサ交換をご依頼ください。

センサ感度の劣化販売店または最寄りの弊社営業所までセンサ交換をご依頼ください。
（電源投入時、測定値に「F A I L 」と表示された場合
は、RESET ボタンを押すことで、警報を解除できます。故障していないセンサを除き、その他のガスについては使用できます。）（VOC センサの場合）
PID ランプが汚れている

販売店または最寄りの弊社営業所まで点検をご依頼ください。

（VOC センサの場合）
電極パレットの劣化販売店または最寄りの弊社営業所まで電極パレッ
トの交換をご依頼ください。

（VOC センサの場合）
PID ランプの劣化販売店または最寄りの弊社営業所まで PID ランプの交換をご依頼ください。

本体回路に異常がある

エラー番号 000

内部の ROM の異常

エラー番号 010

内部の RAM の異常

エラー番号 021

内部の FRAM の異常

エラー番号 031

内部の FLASH メモリの異常販売店または最寄りの弊社営業所まで修理をご依頼ください。

時計異常＜FAIL CLOCK＞

内部の時計異常

日時設定を行ってください。（P.76）
頻繁にこのような症状が起こる場合は、内部時計の故障が考えられますので、交換する必要があります。弊社営業所までご連絡ください。

ユーザーモードに入れない

ユーザーモードのパスワードを忘れてしまった

販売店または最寄りの弊社営業所までご連絡ください。

- 90 -

9 トラブルシューティング

9-2. 指示値の異常

9-2. 指示値の異常症

状

指示値が上がった（下がった）まま元に戻らない

ガス漏れなど、検知ポ
イントに異常がないのにガス警報を発
報する

応答が遅い

ゼロ校正後、ガス漏れなど、検知ポイントに異常がないのに VOC
濃度値が上昇する

VOC センサの感度が著しく低下した

VOC センサの濃度表
示部に「－－－－」が表示され、ランプが点滅しブザーが鳴る

原

因

処

置

センサのドリフト

ゼロ校正を行ってください。（P.38）

干渉ガスの存在

溶剤などの干渉ガスによる影響を完全になくすことは困難です。除去フィルタなどの対策については、販売店または最寄りの弊社営業所までご連絡ください。
検知対象ガスが微量に漏れている可能性があります（スローリーク）。放置しておくと危険な状態になる可能性がありますので、ガス警報時の対応と同等の対応と処置を施してください。

スローリーク

環境の変化

ゼロ校正を行ってください。（P.39）
特に、ガルバニ電池式は気圧の影響を受けます。

干渉ガスの存在

溶剤などの干渉ガスによる影響を完全になくすことは困難です。除去フィルタなどの対策については、販売店または最寄りの弊社営業所までご連絡ください。

ノイズの影響いったん電源を切り、再度電源を入れて（再起動）く
ださい。
頻繁にこのような症状が起こる場合は、適切なノイズ対策を実施してください。

ダストフィルタの詰まり

ダストフィルタを交換してください。（P.85）

テーパーノズルの折れ、詰まり

不具合の箇所を修復してください。

本器内で結露が発生している

ドライエアを導入するなど、不具合の箇所を修復してください。

センサ感度の劣化販売店または最寄りの弊社営業所までセンサ交換をご依頼ください。

電極パレットの劣化販売店または最寄りの弊社営業所まで電極パレッ
トの交換をご依頼ください。

PID ランプが汚れている

販売店または最寄りの弊社営業所まで点検をご依頼ください。

PID ランプの劣化販売店または最寄りの弊社営業所までランプの交換をご依頼ください。

高濃度のマイナス干渉ガス（CH4 など）の存在

新鮮な空気を吸わせてください。ガスの影響が無くなると、10 秒程度で自動的に測定を再開します。

PID ランプの汚れ

新鮮な空気を吸わせてください。新鮮な空気を吸わせても復帰しない場合は、再起動してください。再起動後に「SENSOR FAIL」が表示される場合は、センサが劣化している恐れがあります。販売店または最寄りの弊社営業所まで交換をご依頼ください。

PID ランプの劣化
電極パレットの劣化- 91 -

10 製品仕様

10-1. 仕様一覧

10
10

製品仕様

10-1. 仕様一覧
＜共通仕様＞
濃度表示

LCD デジタル（フルドットディスプレイ、160×128 ドット）検知方式

ポンプ吸引式

吸引流量

0.45L/min 以上（オープン流量）

各種表示

時計表示/電池残量表示/動作状態表示/フロー確認表示

表示言語

日本語/英語/フランス語/スペイン語/ポルトガル語/ドイツ語/イタリア語/ロシア語/韓国語

ブザー音量

95dB（A）以上（30cm）（プロテクトカバー装着時）

ガス警報表示

ランプ点滅/ブザー連続変調鳴動/ガス濃度表示および警報内容点滅/振動

ガス警報動作自己保持

故障警報・自己診断

システム異常/センサ異常/電池電圧低下/校正不良/流量低下

故障警報表示

ランプ点滅/ブザー断続/内容表示

故障警報動作自己保持

パニック警報表示

予備警報：ランプ点滅/ブザー断続（予備警報）
本警報：ランプ点滅/ブザー連続変調鳴動

パニック警報動作

自己保持

マンダウン警報表示※1

予備警報：ランプ点滅/ブザー断続（予備警報）
本警報：ランプ点滅/ブザー連続変調鳴動

マンダウン警報動作※1

自動復帰

伝送仕様

IrDA（データロガ用）

電源

専用リチウムイオン電池ユニット【BUL-6000】※2 または専用乾電池ユニット＜単 3 形アルカリ乾電池×3 本＞【BUD-6000】

連続使用時間

BUL-6000 ：約 14 時間（25℃・無警報・無照明時）
BUD-6000 ：約 8 時間（25℃・無警報・無照明時）

使用温度範囲－20～＋50℃（急変なきこと）

使用湿度範囲

95%RH 以下（結露なきこと）

構造

防塵防滴構造（IP67 相当）（配管除く）

防爆構造

本質安全防爆構造

防爆等級

防爆構造電気機械器具型式検定（国内防爆）： Ex ia IIC T4 X
ATEX/UKEX ：II 1 G Ex ia IIB T4 Ga (可燃性ガスセンサあり、BUL-6000)
II 1 G Ex ia IIC T4 Ga (可燃性ガスセンサなし、BUL-6000)
II 1 G Ex ia IIB T4 Ga (可燃性ガスセンサあり、BUD-6000 東芝)
II 1 G Ex ia IIC T4 Ga (可燃性ガスセンサなし、BUD-6000 東芝)
II 1 G Ex ia IIB T3 Ga (可燃性ガスセンサあり、BUD-6000DURACELL)
II 1 G Ex ia IIC T3 Ga (可燃性ガスセンサなし、BUD-6000DURACELL)- 92 -

10 製品仕様

10-1. 仕様一覧
IECEx ：
Ex ia IIB T4 Ga (可燃性ガスセンサあり、BUL-6000)
Ex ia IIC T4 Ga (可燃性ガスセンサなし、BUL-6000)
Ex ia IIB T4 Ga (可燃性ガスセンサあり、BUD-6000 東芝)
Ex ia IIC T4 Ga (可燃性ガスセンサなし、BUD-6000 東芝)
Ex ia IIB T3 Ga (可燃性ガスセンサあり、BUD-6000DURACELL)
Ex ia IIC T3 Ga (可燃性ガスセンサなし、BUD-6000DURACELL)

外形寸法

約 70（W）×201（H）×54（D）mm（突起部は除く）

質量

約 500g（BUL-6000 使用時）/約 450g（BUD-6000 使用時）

※1 マンダウン警報機能は通常 OFF 設定で使用できません。本機能をご利用になる場合には、販売店または最寄りの弊社営業所までご連絡ください。
※2 JG(国土交通省)型式承認は、充電池仕様(BUL)のみ対応しています。

- 93 -

10 製品仕様

10-1. 仕様一覧

＜センサ別仕様＞
可燃性（HC/CH4）※1<%LEL>

酸素
（O2）

硫化水素
（H2S）

一酸化炭素
（CO）

ニューセラミック式

ガルバニ電池式

定電位電解式

定電位電解式0～100%LEL

0～25.0%＜～40.0vol%＞0～30.0ppm

0～150ppm＜～500ppm＞

最小分解能

1%LEL

0.1vol%0.1ppm

1ppm

警報設定値

10%LEL（AL1）
50%LEL（AL2）
100%LEL（OVER）

19.5vol%（AL1）
18.0vol%（AL2）
40.0vol%（OVER）

1.0ppm（AL1）
10.0ppm（AL2）
1.0ppm（TWA）
5.0ppm（STEL）
30.0ppm（OVER）

25ppm（AL1）
50ppm（AL2）
25ppm（TWA）
200ppm（STEL）
500ppm（OVER）

検知対象ガス

揮発性有機化合物（VOC）＜ppb＞

揮発性有機化合物（VOC）＜ppm＞

二酸化硫黄
（SO2）

二酸化窒素
（NO2）

定電位電解式

定電位電解式

検知対象ガス
検知原理
検知範囲＜サービスレンジ＞

検知原理

光イオン化式（10.6eV） 光イオン化式（10.6eV）

検知範囲

0～40000ppb

0～4000ppm

0～99.90ppm

0～20.00ppm

最小分解能

1ppb
（0～4000ppb）
10ppb（4000～40000ppb）

0.1ppm（0～400.0ppm）
1ppm（400～4000ppm）

0.05ppm

0.05ppm

警報設定値

5000ppb（AL1）
10000ppb（AL2）
40000ppb（OVER）

400.0ppm（AL1）
1000ppm（AL2）4000ppm（OVER）

2.00ppm（AL1）
5.00ppm（AL2）2.00ppm（TWA）
5.00ppm（STEL）
99.90ppm（OVER）

3.00ppm（AL1）
6.00ppm（AL2）
3.00ppm（TWA）
20.00ppm（OVER）

検知対象ガス

シアン化水素
（HCN）

アンモニア
（NH3）

塩素
（Cl2）

ホスフィン
（PH3）

検知原理

定電位電解式

定電位電解式

定電位電解式

定電位電解式

0～15.0ppm

0～400.0ppm

0～10.00ppm

TLV-

最小分解能

0.1ppm

0.5ppm

0.05ppm

0.01ppm

警報設定値

5.0ppm（AL1）
10.0ppm（AL2）
4.7ppm（STEL）
15.0ppm（OVER）

25.0ppm（AL1）
50.0ppm（AL2）
25.0ppm（TWA）
35.0ppm（STEL）
400.0ppm（OVER）

0.50ppm（AL1）
1.00ppm（AL2）0.50ppm（TWA）
1.00ppm（STEL）
10.00ppm（OVER）

0.30ppm (AL1)
1.00ppm(AL2)0.30ppm(TWA)
1.00ppm(STEL)
20.00ppm(OVER)

検知対象ガス

二酸化炭素
（CO2）

二酸化炭素
（CO2）

可燃性ガス（HC）＜%LEL / vol%＞可燃性ガス（CH4）＜%LEL / vol%＞

検知原理

赤外線式

赤外線式

赤外線式

赤外線式

検知範囲

0～10.00vol%0～10000ppm

0～100%LEL＜～30.0vol%＞※20～100%LEL /～100vol%※2

最小分解能

0.02vol%

20ppm

1%LEL / 0.5vol%

1%LEL / 0.5vol%

検知範囲<サービスレンジ>

0.50vol%（AL1）
3.00vol%（AL2）
5000ppm（AL1）
10%LEL / ―（AL1）
10%LEL / ―（AL1）
5000ppm（TWA）
50%LEL / ―（AL2）
50%LEL / ―（AL2）
0.50vol%（TWA）
警報設定値
3.00vol%（STEL）
10000ppm（OVER） 30vol%(OVER)
100.0vol%(OVER)
10.00vol%（OVER）※1 他ガスへの読み替えは、補正係数表を参照ください。また、出荷時 CH4 or HC のいずれかが設定されています(注文時指定)。
※2 100%LEL 以上のガスを検知すると、自動で表示が vol%に切り替わります。

- 94 -

10 製品仕様
検知対象ガス
検知原理

10-1. 仕様一覧
揮発性有機化合物（VOC）
光イオン化式（10.0eV）

測定モード

ノーマルモード

ベンゼンセレクトモード

検知範囲0～100ppm

0～50ppm

0.01ppm（0～10ppm）
0.1ppm（10～100ppm）
5ppm（AL1）
10ppm（AL2）
100ppm（OVER）

0.01ppm（0～10ppm）
0.1ppm（10～50ppm）

最小分解能

警報設定値

50ppm（OVER）- 95 -

10 製品仕様

10-2. 付属品一覧

10-2. 付属品一覧
・リチウムイオン電池ユニット（BUL-6000） / 充電器（1 個）または・乾電池ユニット（BUD-6000） / 単 3 形アルカリ乾電池（3 本）標準付属品オプション品（別売）・プロテクトカバー（1 個）・ベルトクリップ（1 個）・テーパーノズル（1 本）・ハンドストラップ（1 本）・LCD 保護フィルム（1 枚）・活性炭フィルタ（1 個）CF-8350（VOC センサ搭載時のみ付属）
または CF-8501（VOC センサおよび CO2 センサ搭載時のみ付属）
・CO2 除去フィルタ（CF-284）（1 個）（CO2 センサ搭載時のみ付属）
・プレフィルターチューブ（CF-8338）（1 箱 10 本入り）
（VOC<10.0eV>センサ搭載時のみ付属）・チューブホルダー（GF-284）（1 個）（VOC<10.0eV>センサ搭載時のみ付属）
・取扱説明書・製品保証書・リチウムイオン電池ユニット（BUL-6000）
・充電器（1 個）
・乾電池ユニット（BUD-6000）
・単 3 形アルカリ乾電池（3 本）
・ガス採集棒・ガス採集チューブ（0.75m）
・ガス採集チューブ（5m）・ガス採集チューブ（10m）・ガス採集チューブ（20m）
・ガス採集チューブ（30m）・各種フィルタ・PID リスト設定プログラム
・データロガマネジメントプログラム

注意

特定の毒性ガスや溶剤系 VOC など、吸着性の高いガスを測定する際にオプションのガス採集チューブ（0.75m/5m/10m/20m/30m）を使用した場合、チューブ内にガスが吸着し、測定ポイントに実際に存在する検知対象ガス濃度よりも低く指示する恐れがありますので、注意してください。

- 96 -

11 付録11-1. 校正履歴機能／各種トレンド機能／イベント履歴機能

11
11

付録

11-1. 校正履歴機能／各種トレンド機能／イベント履歴機能
本器には履歴・トレンド機能があります。本機能をご使用になる場合は、弊社営業所までお問い合わせください。

注記 履歴／トレンド機能をご使用になる場合は、データロガマネジメントプログラム（別売）が必要です。詳しくは、
弊社営業所までお問い合わせください。

データロガの機能は 5 種類あります。

（1） インターバルトレンド
電源を入れてから切るまでの測定濃度の変化を記録します。
記録数は、最新 3600 回分のデータを記録します。
3600 回を超えると、最古のデータを削除して最新のデータを記録していきます。
※ただし、3600 回以内でも最大記録時間を超えた場合は、最古のデータを削除していきます。
インターバル時間に対しての最大記録時間は次のようになります。
インターバル時間
10 秒
20 秒
30 秒
1分
最大記録時間
10 時間
20 時間※標準のインターバル時間は、「5 分」です。

30 時間

60 時間

3分

5分

10 分

180 時間300 時間600 時間インターバル時間は、「データロガマネジメントプログラム」（別売）で設定できます。

（2） アラームトレンド
警報発報と同時に、発報時間を中心に前後 30 分間（計：1 時間）の測定濃度値の変化を記録します。
アラームトレンドでは、5 秒周期ごとの 5 秒間のピーク値を取得して記録します。
記録数は、最新 8 件分のデータを記録します。
8 件を超えた場合は、最古のデータを削除して最新のデータを記録していきます。

（3） アラームイベント
警報を発報したことを、イベントとして記録します。
記録は、警報発報時間と対象の測定ガスと警報イベントの種類（AL1、AL2、オーバースケール）を記録します。
記録数は、最新のイベントから数えて過去最大 100 件分を記録します。
100 件を超えた場合は、最古のデータを削除して最新データを記録していきます。

（4） 故障イベント
故障を発報したことを、イベントとして記録します。
記録は、故障発報時間と対象の測定ガスや本体機器情報と故障イベントの種類を記録します。
記録数は、最新のイベントから数えて過去最大 100 件分を記録します。
100 件を超えた場合は、最古のデータを削除して最新のデータを記録しています。

- 97 -

11 付録11-1. 校正履歴機能／各種トレンド機能／イベント履歴機能（5） 校正履歴校正を実施する際にデータを記録します。
記録は、校正時間と校正前後の濃度値および校正エラーについても記録します。
記録数は、最新の校正履歴から数えて過去 100 回分のデータを保存します。
100 回を超えた場合は、最古のデータを削除して最新のデータを記録していきます。

注記
 本器のデータロガ機能は、全てオーバーライト（最古のデータを削除して最新のデータを記録する）方式です。
 記録されたデータは、「データロガマネジメントプログラム」（別売）により読み出すことができます。詳細は、「データロガマネジメントプログラム」の取扱説明書を参照してください。

- 98 -

11 付録11-2. 用語の定義

11-2. 用語の定義
ppb

ガス濃度を体積の十億分の 1 の単位で表したものです。

ppm

ガス濃度を体積の百万分の 1 の単位で表したものです。

vol%

ガス濃度を体積の百分の 1 の単位で表したものです。

LEL

爆発下限界の対応英語｢Lower Explosive Limit｣の略語。
爆発下限界とは可燃性ガスが空気と混合して、着火によって爆発を起こす最低濃度。

TWA（時間荷重平均値被曝限度値）｢TLV-TWA：Threshold Limit Value Time Weighted Average｣の略語。1 日 8 時間または週 40 時間の平常作業において、反復被曝しても、ほとんど全ての作業者が健康上悪影響をこうむることがないと考えられる有害物質の時間荷重平均濃度。

STEL（短時間被曝限度値）｢TLV-STEL：Threshold Limit Value Short Term Exposure Limit｣の略語。作業者が 15分間連続被曝しても、かつ、毎日の被曝が TWA 以下あれば健康上作業者に悪影響を及ぼさない有害物質の濃度。

自己保持

警報動作の 1 つで、警報を発した後、警報条件を満たさない場合でも、解除するまで警報を発し続ける動作です。

自動復帰

警報動作の 1 つで、警報を発した後、警報条件を満たさない場合に自動で警報を停止する動作です。

- 99 -

11 付録11-3. VOC 読み替えガスリスト

11-3. VOC 読み替えガスリスト
通常、揮発性有機化合物（VOC）の濃度表示はイソブチレンですが、あらかじめ登録してあるガスに読み替え、濃度表示することができます。設定方法については「PID ガス選択設定」（P.61）を参照してください。
換算係数（10.0eV）の欄に「-」と記載されているガス種については、VOC<10.0eV>センサでは測定できません。

ガス名（表示名）

ガス名（和名）

分子式

CAS 番号換算
係数(10.6eV)

換算
係数(10.0eV)A
Acetaldehyde

アセトアルデヒド

C2H4O

75-07-0

3.4-

Acetamide

アセトアミドC2H5NO

60-35-52-

Acetic acid

酢酸

C2H4O264-19-7

36.2-

Acetic anhydride

無水酢酸

C4H6O3108-24-7

4-Acetoin

アセトイン

C4H8O2513-86-0

1-

Acetone

アセトン

C3H6O

67-64-1

0.7

1.20

Acetophenone

アセトフェノン

C8H8O

98-86-2

0.6-

Acetyl bromide

アセチルブロマイド

C2H3BrO

506-96-7

3-

Acetylglycine, N-

N-アセチルグリシン

C4H7NO3

543-24-8

2-

Acrolein

アクロレイン

C3H4O

107-02-8

3.2-

Acrylic Acid

アクリル酸

C3H4O279-10-7

2.7-

Alkanes, n-, C6+

ノルマルアルカン（C6 以上）

CnH2n+21-

Allyl acetoacetate

アセト酢酸アリル

C7H10O31118-84-9

1.5-Allyl alcohol

アリルアルコール

C3H6O

107-18-6

2.1

4Allyl bromide

アリルブロマイド

C3H5Br

106-95-63-Allyl chloride

アリルクロライド

C3H5Cl

107-05-1

4.5-Allyl glycidyl ether

アリルグリシジルエーテル

C6H10O2

106-92-30.8-Allyl propyl disulfide

アリルプロピルジスルフィド

C6H12S2

2179-59-1

0.4-

Ammonia

アンモニア

NH3

7664-41-7

8.5-

Amyl acetate

酢酸 n-アミル

C7H14O2

628-63-7

1.8

9Amyl alcohol

アミルアルコールC5H12O

71-41-03.5

10Amyl alcohol, tert-

ターシャルアミルアルコール

C5H12O

75-85-4

1.5

2.8

Anethole

アネトール

C10H12O

104-46-1

0.4-

Aniline

アニリン

C6H7N

62-53-3

0.48

0.8

Anisole

アニソール

C7H8O

100-66-3

0.5

0.59

Anisyl aldehyde

アニスアルデヒド

C8H8O2

123-11-5

0.4-

Arsineアルシン

AsH37784-42-1

2.5-

Asphalt, petroleum fumes

天然アスファルタム

8052-42-4

1-

100-52-7

0.9

0.9

B
Benzaldehyde

ベンズアルデヒド- 100 -

C7H6O

11 付録11-3. VOC 読み替えガスリスト

ガス名（表示名）

ガス名（和名）

分子式

CAS 番号換算
係数(10.6eV)

換算
係数(10.0eV)

Benzene

ベンゼン

C6H6

71-43-2

0.46

0.54

Benzene thiol

ベンゼンチオールC6H5SH

108-98-5

0.7

0.8

Benzoic acid

安息香酸

C7H6O265-85-0

0.7-

Benzonitrile

ベンゾニトリル

C7H5N

100-47-0

0.7

0.8

Benzoquinone, o-

オルトベンゾキノン

C6H4O2583-63-1

1-

Benzoquinone, p-

パラベンゾキノン

C6H4O2106-51-4

1-

Benzoyl bromide

ベンゾイルブロマイド

C7H5BrO

618-32-6

2-

Benzyl 2-phenylacetate

フェニル酢酸ベンジル

C15H14O2

102-16-9

0.5-

Benzyl acetate

酢酸ベンジルC9H10O2

140-11-4

0.6-

Benzyl alcohol

ベンジルアルコール

C7H8O

100-51-6

1.3

1.6

Benzyl chloride

塩化ベンジルC7H7Cl100-44-7

0.48

0.7

Benzyl formate

ギ酸ベンジルC8H8O2

104-57-4

0.8-

Benzyl isobutyrate

イソ酪酸ベンジル

C11H14O2

103-28-6

0.5-

Benzyl nitrile

ベンジルシアニド

C8H7N

140-29-4

1-

Benzyl propionate

プロピオン酸ベンジル

C10H12O2

122-63-4

0.5-

Benzylamine

ベンジルアミン

C7H9N

100-46-9

0.6-

Biphenyl

ビフェニル

C12H1092-52-4

0.4

0.6

Borneol

ボルネオール

C10H18O

507-70-0

0.8-

Bromine臭素

Br27726-95-6

15-

Bromo-2,2-dimethylpropane, 1-

1-ブロモ-2,2-ジメチルプロパン

C5H11Br

630-17-1

2-

Bromo-2-chloroethane, 1-

1-ブロモ-2-クロロエタン

C2H4BrCl

107-04-0

8-

Bromo-2-methylpentane, 1-

1-ブロモ-2-メチルペンタン

C6H13Br25346-33-2

2-

Bromoacetone

ブロモアセトン

C3H5BrO

598-31-2

1-Bromoacetylene

ブロモアセチレン

C2HBr

593-61-3

4-

Bromobenzene

ブロモベンゼンC6H5Br108-86-1

0.3

0.32

Bromobutane, 1-

1-ブロモブタン

C4H9Br109-65-9

1

14

Bromobutane, 2-

臭化 sec-ブチル

C4H9Br

78-76-2

1.5

1.6

Bromocyclohexane

ブロモシクロヘキサン

C6H11Br

108-85-0

3-Bromoethane

エチルブロマイド

C2H5Br

74-96-4

5-

Bromoethanol, 2-エチレンブロモヒドリン

C2H5BrO

540-51-2

2-

Bromoethyl methyl ether, 2-2-ブロモエチルメチルエーテル

C3H7BrO

6482-24-2

2.5-

Bromoform

ブロモホルムCHBr3

75-25-2

2.8-

Bromopentane, 1-

1-ブロモペンタン

C5H11Br

110-53-2

2

3.5

Bromopropane, 1-

ノルマルプロピルブロマイド

C3H7Br

106-94-5

1.3

70- 101 -

11 付録11-3. VOC 読み替えガスリスト

ガス名（表示名）

ガス名（和名）

分子式

CAS 番号換算
係数(10.6eV)

換算
係数(10.0eV)

Bromopyridine, 3-3-ブロモピリジン

C5H4BrN

626-55-1

2-

Bromopyridine, 4-4-ブロモピリジン

C5H4BrN

1120-87-2

2-

Bromotrimethylsilane

ブロモトリメチルシラン

C3H9BrSi

2857-97-8

2-

But-2-ynal

2-ブチナール

C4H4O

1119-19-3

3-

But-3-ynal

3-ブチン-1-オン

C4H4O

52844-23-2

1.5-

Butadiene diepoxide, 1,3-

1,3-ブタジエンジエポキシド

C4H6O2

1464-53-5

4-

Butadiene, 1,3-

ブタジエン

C4H6

106-99-00.8

0.8

Butane, n-

ノルマルブタンC4H10

106-97-8

44-

Butanedione, 2,3-

2,3-ブタジオン

C4H6O2431-03-8

0.4

0.87

Butanoic acid

酪酸

C4H8O2

107-92-6

5-

Butanol, 1-

ノルマルブチルアルコール

C4H10O

71-36-3

4

25

Butanol, 2-2-ブチルアルコール

C4H10O

78-92-2

3.0

8

Buten-3-ol, 1-

1-ブテン-3-オール

C4H8O

598-32-3

1.2

3

Butene, 1-

1-ブテン

C4H8

106-98-9

1.5-

Butene, 2-

2-ブテン

C4H8

107-01-7

1.3-

Butene, cis-2-

シス-2-ブテン

C4H8590-18-1

1.3-

Butene, trans-2-

トランス-2-ブテン

C4H8

624-64-6

1.3-

Butenoic acid, 3-

クロトン酸

C4H6O2

107-93-7

2-

Butoxyethanol, 2-エチレングリコールモノブチルエーテル

C6H14O2

111-76-2

1.1-

112-34-5

1.0-

112-07-2

3-

Butoxyethoxyethanol
Butoxyethylacetate, 2-ジエチレングリコールモノブチ
C8H18O3
ルエーテル
エチレングリコールモノブチル
C8H16O3
エーテルアセテート

Butyl acetate

酢酸ブチル

C6H12O2

123-86-4

2.4

12Butyl acetate, sec-

酢酸 sec-ブチル

C6H12O2

105-46-4

2.4

5.5

Butyl acetate, tert-

酢酸ターシャルブチル

C6H12O2

540-88-521.65

Butyl acrylate

アクリル酸ノルマルブチル

C7H12O2

141-32-21.5-Butyl butyrate

酪酸ブチル

C8H16O2

109-21-7

1.8-Butyl chloroformate

クロロギ酸ブチル

C5H9ClO2

592-34-7

3.2-

Butyl cyclohexan-1-ol, 4- tert-

4-ターシャルブチルシクロヘキサン-1-オール

C10H20O

98-52-21.4-Butyl cyclohexyl acetate, 2- tert- 酢酸 2-ターシャルブチルシクロへキシル

C12H22O2

88-41-50.8-Butyl ether, n-

ジブチルエーテル

C8H18O

142-96-1

0.7

1.10

Butyl glycidyl ether

ブチルグリシジルエーテル

C7H14O22426-08-62-

Butyl iodide

ヨウ化ブチル

C4H9I

542-69-81-Butyl isocyanate

ノルマルブチルイソシアネート

C5H9NO

111-36-4

2.5-Butyl lactate

乳酸ブチル

C7H14O3

138-22-7

2.5-- 102 -

11 付録11-3. VOC 読み替えガスリスト

ガス名（表示名）

ガス名（和名）

分子式

CAS 番号換算
係数(10.6eV)

換算
係数(10.0eV)

Butyl mercaptann-ブチルメルカプタン

C4H10S

109-79-50.5-Butyl mercaptan, tert-

ターシャルブチルメルカプタン

C4H10S

75-66-10.4-Butyl methacrylate

メタクリル酸ノルマルブチル

C8H14O2

97-88-11-Butyl propionate, n-

プロピオン酸ブチル

C7H14O2

590-01-2

1.84Butylamine, n-

モノブチルアミン

C4H11N

109-73-91-Butylamine, sec-

sec-ブチルアミン

C4H11N

513-49-50.9-Butylamine, tert-

ターシャルブチルアミン

C4H11N

75-64-90.91.5Butylbenzene

ブチルベンゼンC10H14

104-51-8

0.5

0.45

Butylbenzene, sec-

sec-ブチルベンゼン

C10H14135-98-80.4

0.4

Butylbenzene, tert-

ターシャルブチルベンゼン

C10H1498-06-60.4

0.4

Butylene carbonate, 1,2-

炭酸 1,2-ブチレン

C5H8O34437-85-82-Butylphenol, o-sec-

オルト-sec-ブチルフェノール

C10H14O

89-72-5

0.9-

Butyn-1-ol, 2-2-ブチン-1-オール

C4H6O

764-01-2

1.5-

Butyn-2-one

3-ブチン-2-オン

C4H4O

1423-60-5

3-Butyraldehyde

ノルマルブチルアルデヒド

C4H8O

123-72-8

1.6

1.9

Butyrolactone, gamma-

γ-ブチロラクトン

C4H6O296-48-0

15-

Butyryl chloride

ブチリルクロライド

C4H7ClO

141-75-3

3-Camphene（±）カンフェン

C10H16565-00-4

0.5

0.4

Camphor（±）カンファ―C10H16O

76-22-2

0.4-

Carbon disulfide

二硫化炭素

CS2

75-15-0

1.4

1.3

Carbon suboxide

亜酸化炭素

C3O2

504-64-3

10-

Carbon tetrabromide

四臭化炭素

CBr4

558-13-4

3-

Carene（+）-3-カレン

C10H16

13466-78-9

0.5-

Carvacrol

クルバクロール

C10H14O

499-75-2

0.8-

Carvone, R-ℓ-カルボン

C10H14O

6485-40-1

1

1.5

Caryophyllene(Z)-4,11,11-トリメチル-8-メチレン
C15H24ビシクロ[7.2.0]ウンデカ-4-エン

13877-93-5

0.4-

Chloramine

クロロアミン

ClH2N

10599-90-3

2-Chloro-1,1-difluoroethene, 2-2-クロロ-1,1-ジフルオロエチレン

C2HClF2

359-10-4

1.5-

Chloro-2-propanone, 1-

クロロアセトンC3H5ClO

78-95-5

1-Chloroacetaldehyde

クロロアセトアルデヒド

C2H3ClO

107-20-0

3-Chlorobenzene

モノクロロベンゼン

C6H5Cl108-90-7

0.36

0.5

Chlorobutane, 1-

塩化 n-ブチル

C4H9Cl109-69-3

10-Chlorobutane, 2-2-クロロブタン

C4H9Cl

78-86-4

8-

Chlorocyclohexane

クロロシクロヘキサン

C6H11Cl542-18-7

4

20C- 103 -

11 付録11-3. VOC 読み替えガスリスト

ガス名（表示名）

ガス名（和名）

分子式

CAS 番号換算
係数(10.6eV)

換算
係数(10.0eV)

Chloroethyl methyl ether, 2-2-クロロエチルメチルエーテル

C3H7ClO

627-42-9

2.6-

Chloromethoxyethane

クロロメトキシエタン

C3H7ClO

3188-13-4

4-

Chloroprene

クロロプレン

C4H5Cl126-99-8

1.3-

Chloropyridine, 2-2-クロロピリジン

C5H4ClN

109-09-1

1-

Chlorostyrene, o-

オルトクロロスチレン

C8H7Cl2039-87-4

0.4-

Chlorotoluene, m-

メタクロロトルエン

C7H7Cl108-41-8

0.5-

Chlorotoluene, o-

クロロトルエンC7H7Cl95-49-8

0.5-

Chlorotoluene, p-

パラクロロトルエン

C7H7Cl

106-43-4

0.39

0.3

Chlorotrifluoroethylene

クロロトリフルオロエチレン

C2ClF379-38-9

1-

Cinnamic aldehyde

ケイ皮アルデヒド

C9H8O

104-55-2

0.4-

Cinnamyl acetate

酢酸トランスシンナミル

C11H12O2

21040-45-9

0.4-

Cinnamyl alcohol

シンナミルアルコール

C9H10O

104-54-1

0.4-

Citralシトラール

C10H16O

5392-40-5

1

3.4

Citronellal

シトロネラール

C10H18O

106-23-0

0.9-

Citronellol(±)シトロネロール

C10H20O

26489-01-0

1-

Citronellol acetate

酢酸シトロネリル

C12H22O2

150-84-5

1.5-

Citronellol formate

ギ酸シトロネリル

C11H20O2

105-85-1

1.5-

Citronellyl isobutyrate

イソ酪酸シトロネリル

C14H26O2

97-89-2

0.9-

Coumarin

クマリン

C9H6O291-64-5

0.4-

Creosote

クレオソート8021-39-4

1.0-

Cresol, m-

メタクレゾール

C7H8O

108-39-4

2.2

1.5

Cresol, o-

オルトクレゾール

C7H8O

95-48-7

1.1

1.5

Cresol, p-

パラクレゾール

C7H8O

106-44-5

1.1

1.5

Cresyl acetate, p-

酢酸パラクレジル

C9H10O2

140-39-6

1-Cresyl ethyl ether, p-

パラクレジルエチルエーテル

C9H12O

622-60-6

0.8-

Cresyl methyl ether

パラクレジルメチルエーテル

C8H10O

104-93-8

0.8-

Crotonaldehyde

クロトンアルデヒド

C4H6O

4170-30-3

1-

Crotonyl alcohol

クロチルアルコール

C4H8O

6117-91-5

0.8-

Cumeneクメン

C9H12

98-82-8

0.32-

Cycloalkanes

シクロアルカン

1.5-

Cyclobutanone

シクロブタノン

C4H6O

1191-95-3

1.2-

Cyclobutene

シクロブテン

C4H6

822-35-5

3-

Cycloheptane

シクロへプタンC7H14

291-64-5

1.1-

Cyclohex-2-enedione, 1,4-2-シクロヘキセン-1,4-ジオン

C6H6O24505-38-8

1-

Cyclohexane

シクロヘキサンC6H12

110-82-7

1.2

3.3- 104 -

11 付録11-3. VOC 読み替えガスリスト

ガス名（表示名）

ガス名（和名）

分子式

CAS 番号換算
係数(10.6eV)

換算
係数(10.0eV)

Cyclohexanethiol

シクロヘキサンチオール

C6H12S

1569-69-3

0.5-

Cyclohexanol

シクロヘキサノール

C6H12O

108-93-0

2.9

2.7

Cyclohexanone

シクロヘキサノン

C6H10O

108-94-1

1.1

1.20

Cyclohexene

シクロヘキセンC6H10

110-83-8

0.8

1.4

Cyclohexyl acetate

酢酸シクロへキシル

C8H14O2

622-45-7

1.2-

Cyclohexylamine

シクロへキシルアミン

C6H13N

108-91-8

10.9

Cyclooctadiene

シクロオクタジエン

C8H1229965-97-7

1-

Cyclopentadiene

シクロペンタジエン

C5H6

542-92-7

0.8-

Cyclopentane

シクロペンタンC5H10

287-92-3

12.0-

Cyclopentanone

シクロペンタノン

C5H8O

120-92-3

0.7

1.0

Cyclopentene

シクロペンテン

C5H8

142-29-0

1.5

140

Cyclopentene-1,3-dione, 4-1,4-シクロペンテン-1,3-ジオン

C5H4O2930-60-9

1-

Cyclopropylamine

シクロプロピルアミン

C3H7N

765-30-0

0.8

1.7

Cymene, p-

パラシメン

C10H1499-87-6

0.35-

Decahydronaphthalene

デカヒドロナフタリン

C10H1891-17-8

0.9-

Decanaln-デカナール

C10H20O

112-31-2

0.9-

Decaneノルマルデカン

C10H22

124-18-5

0.9

4.2

Decyne, 1-

1-デシン

C10H18764-93-2

1.3

0.83

Diacetone alcohol

ジアセトンアルコール

C6H12O2

123-42-2

0.8

0.84

Diazine, 1,2-

1,2-ジアジン

C4H4N2289-80-5

3-

Diazine, 1,3-

1,3-ジアジン

C4H4N2289-95-2

3-

Dibromoacetylene

ジブロモアセチレン

C2Br2624-61-3

1.5-

Dibromochloromethane

ジブロモクロロメタン

CHBr2Cl

124-48-1

10-

Dibromocyclohexane, 1,2-

1,2-ジブロモシクロヘキサン

C6H10Br2

5401-62-7

3-

Dibromocyclopentane

1,2-ジブロモシクロペンタン

C5H8Br2

33547-17-0

3-

Dibromodichloromethane

ジブロモジクロロメタン

CBr2Cl2

594-18-3

4-

Dibromoethane, 1,2-エチレンジブロマイド

C2H4Br2

106-93-4

2-

Dibromoethene, 1,1-

1,1-ジブロモエテン

C2H2Br2

593-92-0

1.5-

Dibromoethene, 1,2-

1,2-ジブロモエチレン

C2H2Br2

540-49-8

1.5-

Dibromomethane

ジブロモメタン

CH2Br274-95-3

1.2-

Dichloro-1,2-difluoroethene, 1,2-

1,2-ジクロロ-1,2-ジフルオロエテン

C2Cl2F2

598-88-92-

Dichloro-1-propene, 2,3-

2,3-ジクロロ-1-プロペン

C3H4Cl2

78-88-6

1.4-

Dichloro-2,2,-difluoroethene, 1,1-

1,1-ジクロロ-2,2-ジフルオロエテン

C2Cl2F2

79-35-6

1-

Dichloroacetylene

ジクロロアセチレン

C2Cl27572-29-4

5-D- 105 -

11 付録11-3. VOC 読み替えガスリスト

ガス名（和名）

分子式

CAS 番号換算
係数(10.6eV)

Dichlorobenzene, o-

オルトジクロロベンゼン

C6H4Cl2

95-50-1

0.5

0.5

Dichlorobenzene, p-

パラジクロロベンゼン

C6H4Cl2

106-46-7

0.5

0.5

Dichloroethene, 1,1-

1,1-ジクロロエチレン

C2H2Cl275-35-4

1-

Dichloroethene, 1,2-

1,2-ジクロロエチレン

C2H2Cl2

540-59-0

0.36

0.29

Dichloroethene, cis-1,2-

シス 1,2-ジクロロエチレン

C2H2Cl2

156-59-2

0.8-

Dichloroethene, trans-1,2-

トランス-1,2-ジクロロエチレン

C2H2Cl2

156-60-5

0.36-

Dichloromethane

ジクロロメタンCH2Cl275-09-2

39-

Dichloromethylamine

N,N-ジクロロ-N-メチルアミン

CH3Cl2N

7651-91-4

2-

Dicyclohexylamine

ジシクロへキシルアミン

C12H23N

101-83-7

0.8-

Dicyclopentadiene

ジシクロペンタジエン

C10H1277-73-6

0.9-

Diesel fuelディーゼル燃料

68334-30-50.8-Diethoxyethane, 1,1-

アセタール

C6H14O2

105-57-70.9

1.0Diethyl carbonate

ジエチルカーボネート

C5H10O3

105-58-8

1.5-

Diethyl ether

ジエチルエーテル

C4H10O

60-29-70.9-Diethyl maleate

マレイン酸ジエチルエステル

C8H12O4

141-05-92-Diethyl malonate

マロン酸ジエチル

C7H12O4

105-53-3

4.0-

Diethyl phthalate

フタル酸ジエチル

C12H14O4

84-66-2

1-Diethyl sulfate

ジエチル硫酸C4H10SO4

64-67-5

3-Diethyl sulfide

硫化ジエチルC4H10S

352-93-2

0.6

0.5

Diethyl sulfone

ジエチルスルホン

C4H10O2S

597-35-3

2-Diethylacetylene

3-ヘキシン

C6H10

928-49-4

2-Diethylaminopropylamine, 3-3-（ジエチルアミノ）プロピルアミン

C7H18N2

104-78-9

1.2

3Diethylene glycol monoethyl ether

ジエチレングリコールモノエチルエーテル

C6H14O3

111-90-0

0.6-Diethylenetriamine

ジエチレントリアミン

C4H13N3

111-40-0

0.9-Diethylhydroxylamine

ジエチルヒドロキシルアミン

C4H11NO

3710-84-7

2

1.5

Diethylsilane

ジエチルシラン

C4H12Si

542-91-6

2-

Diglycidyl ether

ジグリシジルエーテル

C6H10O3

2238-07-5

3-

Dihydroeugenol

ジヒドロオイゲノール

C10H14O2

2785-87-7

0.4-

Dihydrojasmone

ジヒドロジャスモン

C11H18O

1128-08-1

0.6-

Dihydromyrcenol

ジヒドロミルセノール

C10H20O

18479-58-8

0.8-

Dihydroxybenzene, 1,2-

カテコール

C6H6O2120-80-9

1-Dihydroxybenzene, 1,3-

レゾルシン

C6H6O2

108-46-3

1-

Diiodomethane

ジヨードメタンCH2I2

75-11-6

1.2-

Diisobutyl ketone

ジイソブチルケトン

C9H18O

108-83-8

0.8

0.7

Diisobutylene

2,4,4-トリメチルペンテン-1

C8H16

107-39-10.60.9

ガス名（表示名）- 106 -

換算
係数(10.0eV)

11 付録11-3. VOC 読み替えガスリスト

ガス名（和名）

分子式

CAS 番号換算
係数(10.6eV)

Diisopropyl ether

イソプロピルエーテル

C6H14O108-20-3

0.7

0.95

Diisopropylbenzene

ジイソプロピルベンゼン

C12H1825321-09-9

0.4-

Diketene

ジケテン

C4H4O2674-82-8

2.2-

Dimethoxybenzene, 1,4-

パラジメトキシベンゼン

C8H10O2

150-78-7

1.3-

Dimethoxyethane, 1,2-エチレングリコールジメチルエーテル

C4H10O2

110-71-4

1.2

1.2

Dimethoxymethane

ジメトキシメタン

C3H8O2

109-87-5

1.4

13

Dimethyl carbonate

ジメチルカーボネート

C3H6O3616-38-6

2.0-

Dimethyl disulfide

ジメチルジスルフィド

C2H6S2624-92-0

0.2-

Dimethyl ether

ジメチルエーテル

C2H6O

115-10-6

1.3-

Dimethyl phthalate

フタル酸ジメチル

C10H10O4

131-11-3

1-Dimethyl sulfoxide

ジメチルスルホキシド

C2H6OS

67-68-5

1

32

Dimethylacetamide N,N-

N,N-ジメチルアセトアミド

C4H9NO

127-19-5

1.3-

Dimethylacetylene

2-ブチン

C4H6

503-17-3

1-

Dimethylaminoethanol, 2-

N,N-ジメチルエタノールアミン

C4H11NO

108-01-0

1.5-

Dimethylaniline, NN-

N,N-ジメチルアニリン

C8H11N

121-69-7

0.6

0.5

Dimethylboron bromide

ブロモジメチルボラン

C2H6BBr5158-50-94-Dimethylbutyl acetate

酢酸 4-メチル-2-ペンチル

C8H16O2

108-84-9

1.6-

Dimethylcycloheptane, 1,2-

トランス-1,2-ジメチルシクロへプタン

C9H1813151-50-3

1.3-

Dimethylcyclohexane,1,2-ジメチルシクロヘキサン

C8H16583-57-3

0.8

0.9

Dimethylcyclopentane

1α,2α-ジメチルシクロペンタン

C7H141192-18-3

1.2-

Dimethylethylamine, NN-

N,N-ジメチルエチルアミン

C4H11N

598-56-1

3

1.7

Dimethylformamide

N,N-ジメチルホルムアミド

C3H7NO

68-12-2

0.8

1.1

Dimethylhydrazine, 1,1-ジメチルヒドラジン

C2H8N257-14-7

1-

Dimethyloctan-1-ol, 3,7-

3,7-ジメチル-1-オクタノール

C10H22O

106-21-8

1.2-

Dimethyloctan-3-ol, 3,7-テトラヒドロリナロール

C10H22O

78-69-3

1.2-

Dimethylpentane, 2,4-

2,4-ジメチルペンタン

C7H16

108-08-7

1.0-

Dimethylsilane

ジメチルシランC2H8Si

1111-74-62-

Dimethylthiophosphoryl
chloride

ジメチルチオリン酸クロライド

C2H6ClO2PS

2524-03-0

1-

Di-n-butylamine

ジブチルアミン

C8H19N

111-92-20.9

4

Di-n-propylamine

ジノルマルプロピルアミン

C6H15N

142-84-7

1

1.5

Dioxane, 1,4-

ジオキサン

C4H8O2123-91-1

1.5

1.7

Dioxolane

1,3-ジオキソラン

C3H6O2646-06-0

1.8

4.5

Dipentened-リモネン

C10H16138-86-3

0.9

0.8

Diphenyl ether

ジフェニルエーテル

C12H10O

101-84-8

0.8

1.7

Dipropyl ether

プロピルエーテル

C6H14O

111-43-30.8-

ガス名（表示名）- 107 -

換算
係数(10.0eV)

11 付録11-3. VOC 読み替えガスリスト

ガス名（和名）

分子式

CAS 番号換算
係数(10.6eV)

Dipropylene glycol

ジプロピレングリコール

C6H14O3

110-98-5

4-

Disilane

ジシラン

Si2H6

1590-87-0

2-

Disulfur dibromide

二臭化二硫黄Br2S213172-31-1

1.5-

Di-tert-butyl-p-cresol

2,6-ジ-t-ブチル-p-クレゾール

C15H24O

128-37-0

0.3-

Divinylbenzene

ジビニルベンゼン

C10H101321-74-0

0.4

0.4

Divinylbenzene, 1,3-

1,3-ジビニルベンゼン

C10H10108-57-6

0.3

0.25

Dodecenen-ドデカン

C12H26112-40-3

0.8-

Epichlorohydrin

エピクロロヒドリン

C3H5ClO

106-89-8

3.4

30

Epoxypropyl isopropyl ether, 2,3-グリシジルイソプロピルエーテル

C6H12O2

4016-14-2

1.1

1.1

Estagole

4-アリルアニソール

C10H12O

140-67-0

0.7-Ethanol

エチルアルコール

C2H6O

64-17-5

8.7-

Ethanolamine

モノエタノールアミン

C2H7NO

141-43-5

3-Ethoxy-2-methylpropane, 1-

1-エトキシ-2-メチルプロパン

C6H14O

627-02-1

0.8-

Ethoxy-2-propanol, 1-

プロピレングリコールモノエチルエーテル

C5H12O2

1569-02-4

2-

Ethoxy-butane, 2-2-エトキシ-ブタン

C6H14O

19316-73-5

0.8-

Ethoxyethanol, 2-エチレングリコールモノエチルエーテル

C4H10O2

110-80-52

5

111-15-93-

461-24-55-

ガス名（表示名）換算
係数(10.0eV)E

Ethoxyethyl acetate, 2Ethyl 2,2,2-trifluoroethyl ether

エチレングリコールモノエチル
C6H12O3
エーテルアセテート
2,2,2-トリフルオロエチルエチ
C4H7F3O
ルエーテル

Ethyl 2-methylbutyrate

DL-2-メチル酪酸エチル

C7H14O27452-79-121.8Ethyl acetate

酢酸エチル

C4H8O2141-78-63.640Ethyl acetoacetate

アセト酢酸エチル

C6H10O3

141-97-93-Ethyl acrylate

アクリル酸エチル

C5H8O2140-88-5215Ethyl benzoate

安息香酸エチル

C9H10O2

93-89-00.9-Ethyl butyrate

酪酸エチル

C6H12O2

105-54-413.3Ethyl chloroformate

クロロ炭酸エチルエステル

C3H5O2Cl

541-41-383-Ethyl cyanoacrylate

2-シアノアクリル酸エチル

C6H7O2N

7085-85-01.5-Ethyl decanoate

デカン酸エチル

C12H24O2

110-38-31.8-Ethyl formate

ギ酸エチル

C3H6O2109-94-429.8-

Ethyl hexanoate

カプロン酸エチル

C8H16O2

123-66-0

2.63.3

Ethyl hexanol, 2-2-エチルヘキサノール

C8H18O

104-76-7

1.5-Ethyl hexyl acrylate, 2-

アクリル酸-2-エチルへキシル

C11H20O2

103-11-71-Ethyl iodide

ヨウ化エチル

C2H5I

75-03-61.20.30

Ethyl isopropyl ketone

2-メチル-3-ペンタノン

C6H12O

565-69-5

0.8-- 108 -

11 付録11-3. VOC 読み替えガスリスト

ガス名（表示名）

ガス名（和名）

分子式

CAS 番号換算
係数(10.6eV)

換算
係数(10.0eV)

Ethyl lactate

乳酸エチル

C5H10O3

97-64-335Ethyl mercaptan

エチルメルカプタン

C2H6S

75-08-1

0.56

0.55

Ethyl methacrylate

メタクリル酸エチル

C6H10O2

97-63-21.5

1.6Ethyl methyl carbonate

エチルメチルカーボネート

C4H8O3623-53-0

1.5-Ethyl morpholine, 4-N-エチルモルホリン

C6H13NO

100-74-30.6-Ethyl octanoate

カプリル酸エチル

C10H20O2

106-32-1

2.3-Ethyl phenyl acetate

フェニル酢酸エチル

C10H12O2

101-97-31.2-Ethyl propanoate

プロピオン酸エチル

C5H10O2

105-37-326Ethyl tert-butyl ether

エチルターシャルブチルエーテル

C6H14O

637-92-30.6-Ethyl-2-methyl benzene, 1-2-エチルトルエン

C9H12611-14-3

0.45

0.5

Ethyl-3-ethoxypropionate

エチル-3-エトキシプロピオネート

C7H14O3

763-69-93-Ethylacetylene

エチルアセチレン

C4H6

107-00-63-Ethylamine

モノエチルアミン

C2H7N

75-04-71-Ethylbenzene

エチルベンゼンC8H10

100-41-40.50.6Ethylcyclohexane

エチルシクロヘキサン

C8H16

1678-91-711.3

Ethylene

エチレン

C2H4

74-85-18-Ethylene carbonate

エチレンカーボネート

C3H4O396-49-13-Ethylene glycol

エチレングリコール

C2H6O2

107-21-1209Ethylene glycol diacetate

ジ酢酸エチレングリコール

C6H10O4

111-55-74-Ethyleneether

2-プロポキシエタノール

C5H12O2

2807-30-93-Ethylene oxide

酸化エチレン

C2H4O

75-21-815-Ethylenediamine

エチレンジアミン

C2H8N2

107-15-30.810Ethyleneimine

エチレンイミン

C2H5N

151-56-42-Ethylhexanal, 2-2-エチルヘキサナール

C8H16O

123-05-71.5-Ethylhexanoic acid, 2-2-エチルヘキサン酸

C8H16O2

149-57-52.0

16Ethylhexenal, 2-2-エチルヘキセナール

C8H14O

645-62-5

1.3-

Eucalyptol

1,8-シネオール

C10H18O

470-82-6

0.6-

Eugenol

オイゲノール

C10H12O2

97-53-0

0.4-

Eugenol methyl ether

メチルオイゲノール

C11H14O2

93-15-2

0.4-

Fenchol

フェンキルアルコール

C10H18O

1632-73-1

0.4-

Ferrocene

フェロセン

C10H10Fe

102-54-5

0.8-

Fluorobenzene

フルオロベンゼン

C6H5F

462-06-6

0.8

0.83

Fluorobenzoic acid, 4-

4-フルオロ安息香酸

C7H5FO2

456-22-4

2-Formamide

ホルムアミドCH3ON

75-12-72-glycolmonopropylF- 109 -

11 付録11-3. VOC 読み替えガスリスト

ガス名（表示名）

ガス名（和名）

分子式

CAS 番号換算
係数(10.6eV)

換算
係数(10.0eV)

Furan

フラン

C4H4O

110-00-9

0.4-

Furfural

フルフラールC5H4O298-01-1

0.82-

Furfuryl alcohol

フルフリルアルコール

C5H6O298-00-0

2-

Furfuryl mercaptan

フルフリルメルカプタン

C5H6OS

98-02-20.5-

8006-61-9

0.8

1

G
Gasoline

ガソリン

Geranial

ゲラニアール

C10H16O

141-27-5

0.6-

Geraniol

ゲラニオ―ル

C10H18O

106-24-1

0.7-

Geranyl acetate

酢酸ゲラニルC12H20O2

105-87-3

1.2-

Germane

ゲルマン

GeH4

7782-65-2

10-Glutaraldehyde

グルタルアルデヒド

C5H8O2111-30-8

0.9-

Glycidyl methacrylate

メタクリル酸グリシジル

C7H10O3

106-91-2

1.2-

Glycolaldehyde

ヒドロキシアセトアルデヒド

C2H4O2141-46-8

5.0-

Glyoxal

グリオキザール

C2H2O2

107-22-2

1-

GuaiacolグアヤコールC7H8O290-05-1

0.8-

Heptan-2-one

メチルアミルケトン

C7H14O

110-43-0

0.7

0.97

Heptan-3-one

3-ヘプタノン

C7H14O

106-35-4

0.8

0.81

Heptaneノルマルヘプタン

C7H16

142-82-5

1.6

11

Heptanol

ヘプタノールC7H16O

53535-33-4

1.7-

Heptene, 1-

1-ヘプテン

C7H14

592-76-7

0.9

1.1

Heptylcyclopentan-1-one, 2-2-ヘプチルシクロペンタノン

C12H22O

137-03-1

0.8-

Heptyne, 1-

1-ヘプチン

C7H12

628-71-7

2-

Hex-1-en-3-ol

1-ヘキセン-3-オール

C6H12O

4798-44-1

0.9-

Hexachlorodisilane

ヘキサクロロジシラン

Cl6Si213465-77-5

8-

Hexamethyldisilazane,
1,1,1,3,3,3-.

ヘキサメチルジシラザン

C6H19NSi2

999-97-3

1-

Hexamethyldisiloxane

ヘキサメチルジシロキサン

C6H18OSi2

107-46-0

0.3-

Hexamethylene diisocyanate

ヘキサメチレンジイソシアネート

C8H12N2O2

822-06-0

1.5-

Hexan-2-one

メチルノルマルブチルケトン

C6H12O

591-78-6

0.8

0.7

Hexaneノルマルヘキサン

C6H14

110-54-3

2.6

13

Hexanoic acidn-カプロン酸

C6H12O2

142-62-1

3-

Hexanol

ノルマルへキシルアルコール

C6H14O

111-27-3

2

7Hexene, 1-

1-ヘキセン

C6H12

592-41-6

0.9

1.1

Hexenyl acetate, cis-3-

酢酸シス-3-ヘキセニル

C8H14O23681-71-8

1.5

1.2

Hexenyl butyrate, cis-3-酪酸シス-3-ヘキセニル

C10H18O2

16491-36-4

1.5

-

H

- 110 -

11 付録11-3. VOC 読み替えガスリスト

ガス名（表示名）

ガス名（和名）

分子式

CAS 番号換算
係数(10.6eV)

換算
係数(10.0eV)

Hexylaldehyde

ヘキサナールC6H12O

66-25-1

0.6

1.8

Hydrazine

ヒドラジン

H4N2

302-01-2

3-

Hydrogen iodide

ヨウ化水素

HI

10034-85-2

5-

Hydrogen selenide

セレン化水素H2Se

7783-07-5

2-

Hydrogen sulfide

硫化水素

H2S

7783-06-4

4-

Hydrogen telluride

テルル化水素H2Te7783-09-7

1.5-

Hydroxybutanal, 3-3-ヒドロキシブタナール

C4H8O2

107-89-1

2.0-

Hydroxycitronellal

ヒドロキシシトロネラール

C10H20O2

107-75-5

1-

Hydroxyethyl acrylate

2-ヒドロキシエチルアクリレート

C5H8O3818-61-1

1.2-

Hydroxylamine

ヒドロキシルアミン

H3NO

7803-49-8

2-

Hydroxypropyl acrylate, 2-2-ヒドロキシプロピルアクリレート

C6H10O3

999-61-1

1.5-

Indeneインデン

C9H8

95-13-6

0.5

0.4

Indole

インドール

C8H7N

120-72-9

0.4-

Iodineヨウ素I27553-56-2

0.2

0.1

Iodobenzene

ヨードベンゼン

C6H5I

591-50-4

0.2-

Iodoethene

ヨードエテン

C2H3I

593-66-8

1.2-

Iodoform

ヨードホルムCHI3

75-47-8

1.5-

Iodomethane

ヨウ化メチルCH3I

74-88-4

0.4-

Isoalkanes, C10-C13

イソアルカン （C10-C13）

68551-17-7

1-

Isoamyl acetate

酢酸イソアミル

C7H14O2

123-92-2

1.6

6

Isoamyl salicilate

サリチル酸イソアミル

C12H16O3

87-20-71-

Isoamylene

イソアミレン

C5H10

513-35-910.86

Isobornyl acetate

酢酸イソボルニル

C12H20O2

125-12-2

0.4-

Isobutane

イソブタン

C4H10

75-28-58-Isobutanol

イソブチルアルコール

C4H10O

78-83-1

3.5

13

Isobutyl acetate

酢酸イソブチル

C6H12O2

110-19-0

2.3

10Isobutyl acrylate

イソブチルアクリレート

C7H12O2

106-63-8

1.3

5Isobutylbenzene

イソブチルベンゼン

C10H14538-93-2

0.4

0.4

Isobutylene

イソブチレン

C4H8

115-11-711Isobutylene epoxide

イソブチレンオキシド

C4H8O

558-30-5

3-Isobutyraldehyde

イソブチルアルデヒド

C4H8O

78-84-2

1.2-

Isobutyric acid

イソ酪酸

C4H8O279-31-2

4

15

Isodecanol

イソデシルアルコール

C10H22O

25339-17-7

0.9-

Isoeugenol

イソオイゲノール

C10H12O2

97-54-1

0.4-

I- 111 -

11 付録11-3. VOC 読み替えガスリスト

ガス名（表示名）

ガス名（和名）

分子式

CAS 番号換算
係数(10.6eV)

換算
係数(10.0eV)

Isoheptane

2-メチルヘキサン

C7H16591-76-4

1.2-

Isojasmone

ジヒドロイソジャスモン

C11H18O

95-41-00.7-

Isomenthone（+）-イソメントン

C10H18O

1196-31-2

0.6-

Isononanal

3,5,5-トリメチルヘキサナール

C9H18O

5435-64-3

9.0

1.4

Isononanol

3,5,5-トリメチル-1-ヘキサノール

C9H20O

3452-97-9

1.5-

Isooctane

2,3,4-トリメチルペンタン

C8H18

565-75-3

0.74

3.2

Isooctanol

イソオクタノール

C8H18O

26952-21-6

1.7-

Isopentane

イソペンタン

C5H12

78-78-44.0-

Isopentene

2-メチル-1-ブテン

C5H10

563-46-2

0.8-

Isophorone

イソホロン

C9H14O

78-59-10.8

1.0Isophorone diisocyanate

ジイソシアン酸イソホロン

C12H18N2O2

4098-71-9

0.6-

Isoprene

イソプレン

C5H8

78-79-50.8-

Isopropanol

イソプロピルアルコール

C3H8O

67-63-04.4

25

Isopropanolamine

モノイソプロパノールアミン

C3H9NO

78-96-6

1.5-

Isopropoxyethanol, 2-エチレングリコールモノイソプロピルエーテル

C5H12O2

109-59-1

1.5

1.5

Isopropyl acetate

酢酸イソプロピル

C5H10O2

108-21-4

2.2

8

Isopropyl chloroformate

クロロ炭酸イソプロピルエステル

C4H7O2Cl

108-23-6

1.6-

Isopropyl mercaptan

2-プロパンチオール

C3H8S

75-33-2

0.56-

Isopropyl nitrite

亜硝酸イソプロピル

C3H7NO2

541-42-44-Isopropylamine

イソプロピルアミン

C3H9N

75-31-01.2

1Isopropylaminoethanol, 2-2-イソプロピルアミノエタノール

C5H13NO

109-56-82-Isopropylcyclohexane

イソプロピルシクロヘキサン

C9H18

696-29-70.91.1

Isothiazole

イソチアゾール

C3H3NS

288-16-4

3-Isovaleraldehyde

イソバレルアルデヒド

C5H10O

590-86-3

1.3

1.5

Isovaleric acid

イソ吉草酸

C5H10O2

503-74-2

3.0

25

Isoxazole

イソオキサゾール

C3H3NO

288-14-2

6-

Jasmal

1,3-ノナンジオールアセテート

C11H22O3

1322-17-4

1.4-

Jasmone, cis-

シスジャスモンC11H16O

488-10-8

0.5-

Jet Fuel JP-4

JP-4

0.8

0.7

Jet Fuel JP-5

JP-5

0.7

0.6

Jet Fuel JP-8

JP-8

0.7

0.6

8008-20-6

0.8

0.7

463-51-4

3

-

J

K
Kerosene

ケロシン

Keteneケテン

C2H2O- 112 -

11 付録11-3. VOC 読み替えガスリスト

分子式

CAS 番号換算
係数(10.6eV)

Linalool oxide

2,2,6-トリメチル-6-ビニルテト
C10H18O2
ラヒドロピラン-3-オール

14049-11-7

0.6-

Linalyl acetate

酢酸リナリルC12H20O2

115-95-7

0.9-

Maleic anhydride

無水マレイン酸

C4H2O3108-31-6

2-

Menthol

メントル

C10H20O

1490-04-6

0.5-

Menthone

メントン

C10H18O

89-80-5

0.4-

Mercaptoacetic acid

チオグリコール酸

C2H4O2S

68-11-1

1-

Metaldehyde

メタアルデヒドC8H16O4

108-62-3

2.0-

Methacrylamide

メタクリルアミド

C4H7NO

79-39-0

2.0-

Methacrylic acid

メタクリル酸C4H6O279-41-4

2.3-

Methacrylonitrile

メタクリロニトリル

C4H5N

126-98-7

5-

Methanol

メチルアルコール

CH4O

67-56-1

200-

Methoxy-1-butanol, 3-3-メトキシ-1-ブタノール

C5H12O22517-43-3

3-

Methoxy-1-propanol, 2-2-メトキシ-1-プロパノール

C4H10O2

1589-47-5

2-

Methoxy-2,2-dimethylpropane

1-メトキシ-2,2-ジメチルプロパン

C6H14O

1118-00-9

0.7-

Methoxybutyl acetate, 3-

酢酸メトキシブチル

C7H14O34435-53-4

2-

Methoxyethane

エチルメチルエーテル

C3H8O

540-67-0

1.0-

Methoxyethanol, 2-エチレングリコールモノメチルエーテル

C3H8O2

109-86-4

2.7-

Methoxyethene

ビニルメチルエーテル

C3H6O

107-25-5

1-

Methoxyethoxyethanol, 2-ジエチレングリコールモノメチルエーテル

C5H12O3

111-77-3

1.4-

C5H10O3

110-49-6

2.7-

C6H14O3

111-96-6

0.8-

C7H16O334590-94-8

1.3-

C4H10O2

107-98-222.7

ガス名（表示名）

ガス名（和名）換算
係数(10.0eV)L

M

Methoxyethyl acetate
Methoxyethyl ether, 2Methoxymethylethoxy-2-propanol
Methoxypropan-2-ol, 1-

エチレングリコールモノメチル
エーテルアセテート
ジエチレングリコールジメチル
エーテル
ジプロピレングリコールモノメチ
ルエーテル
プロピレングリコールモノメチ
ルエーテル

Methoxypropane, 2-2-メトキシプロパン

C4H10O

598-53-8

0.9-

Methoxypropyl acetate

プロピレングリコールモノメチ
C6H12O3
ルエーテルアセテート

108-65-6

1.2

2.1

Methyl 2-methylpropanoate

イソ酪酸メチル

C5H10O2

547-63-7

2-

Methyl acetate

酢酸メチル

C3H6O279-20-9

5.2-

Methyl acetoacetate

アセト酢酸メチル

C5H8O3105-45-3

3-

Methyl acrylate

アクリル酸メチル

C4H6O2

96-33-3

3.4

80

Methyl anthranilate

アンスラニル酸メチル

C8H9NO2

134-20-3

0.4-

Methyl benzoate

安息香酸メチル

C8H8O2

93-58-3

1.2-- 113 -

11 付録11-3. VOC 読み替えガスリスト

ガス名（表示名）

ガス名（和名）

分子式

CAS 番号換算
係数(10.6eV)

換算
係数(10.0eV)

Methyl bromide

臭化メチル

CH3Br

74-83-9

1.9-

Methyl dimethylacrylate

3,3-ジメチルアクリル酸メチル

C6H10O2

924-50-5

2.5-

Methyl ethyl ketone

メチルエチルケトン

C4H8O

78-93-3

0.8

2

Methyl ethyl ketone peroxides

メチルエチルケトンパーオキサイド

C8H18O6

1338-23-4

0.8-

Methyl heptyne carbonate

2-オクチン酸メチル

C9H14O2

111-12-6

1.3-

Methyl ionone

メチルヨノン

C14H22O

1335-46-2

0.4-

Methyl isobutyl ketone

メチルイソブチルケトン

C6H12O

108-10-1

0.8

1.01

Methyl isocyanate

メチルイソシアナート

C2H3NO

624-83-9

5-

Methyl isopropyl ketone

メチルイソプロピルケトン

C5H10O

563-80-4

0.8

0.96

Methyl isothiocyanate

メチルイソチオシアネート

C2H3NS

556-61-6

0.6-

Methyl mercaptan

メチルメルカプタン

CH4S

74-93-1

0.7

0.6

Methyl methacrylate

メタクリル酸メチル

C5H8O280-62-6

1.6

2.1

Methyl phenyl acetate

フェニル酢酸メチル

C9H10O2

101-41-7

0.4-

Methyl propargyl ether

メチルプロパルギルエーテル

C4H6O

627-41-8

2-

Methyl propionate

プロピオン酸メチル

C4H8O2554-12-1

1.5

36

Methyl propynoate

プロピオル酸メチル

C4H4O2922-67-8

10-

Methyl salicylate

サリチル酸メチル

C8H8O3

119-36-8

0.8-

Methyl sulfide

硫化ジメチル

C2H6S

75-18-3

0.5

0.7

Methyl tert-butyl ether

メチルターシャルブチルエーテル

C5H12O

1634-04-4

0.8

1.02

Methyl thiocyanate

チオシアン酸メチル

C2H3NS

556-64-9

2-

Methyl thioglyconate

チオグリコール酸メチル

C3H6O2S

2365-48-2

1-

Methyl undecanal, 2-2-メチルウンデカナール

C12H24O

110-41-8

1.1-

Methyl vinyl ketone

メチルビニルケトン

C4H6O

78-94-4

0.6-

Methyl-1-butene, 3-3-メチル-1-ブテン

C5H10

563-45-1

0.8-

Methyl-2-butanol, 3-3-メチル-2-ブタノール

C5H12O

598-75-4

3.3-

Methyl-2-hexenoic acid, trans-3-

トランス-3-メチル-2-ヘキセン酸

C7H12O227960-21-0

1.5-

Methyl-2-propen-1-ol, 2-β-メタリルアルコール

C4H8O

513-42-8

1.1

1.6

Methyl-2-pyrrolidinone, N-

N-メチル-2-ピロリドン

C5H9NO

872-50-4

0.9-

Methyl-5-hepten-2-one, 6-6-メチル-5-ヘプテン-2-オン

C8H14O

110-93-0

0.8

0.76

Methylamine

モノメチルアミン

CH5N

74-89-5

1.4-

Methylbutan-1-ol, 3-

イソアミルアルコール

C5H12O

123-51-3

3

10

Methylbutanal, 2-2-メチルブチルアルデヒド

C5H10O

96-17-3

1.5

1.3

Methylbutanol

2-メチルブタノール

C5H12O

137-32-6

1.5-

Methylbutyric acid, 2-

DL-2-メチル酪酸

C5H10O2

116-53-0

3.5

20

Methylcyclohexane

メチルシクロヘキサン

C7H14

108-87-2

1.1

1- 114 -

11 付録11-3. VOC 読み替えガスリスト

ガス名（表示名）

ガス名（和名）

分子式

CAS 番号換算
係数(10.6eV)

換算
係数(10.0eV)

Methylcyclohexanol

メチルシクロヘキサノール

C7H14O

25639-42-3

2.4-

Methylcyclohexanol, 4-4-メチルシクロヘキサノール

C7H14O

589-91-3

2.4-

Methylcyclohexanone, 2-2-メチルシクロヘキサノン

C7H12O

583-60-8

1-Methylcyclopentane

メチルシクロペンタン

C6H12

96-37-7

1.5-

Methylenepentane, 3-2-エチル-1-ブテン

C6H12760-21-4

0.8-

Methylheptan-3-one, 5-5-メチル-3-ヘプタノン

C8H16O

541-85-5

0.8

0.88

Methylhexan-2-one, 5-メチルイソアミルケトン

C7H14O

110-12-3

0.8

0.91

Methylhydrazine

モノメチルヒドラジン

CH6N260-34-4

1.3-

Methylpent-3-en-2-one, 4-

メシチルオキサイド

C6H10O

141-79-7

0.7

0.66

Methylpentan-2-ol, 4-4-メチル-2-ペンタノール

C6H14O

108-11-2

2.8

3

Methylpentane, 2-2-メチルペンタン

C6H14

107-83-5

1.5

34

Methylpentane, 3-3-メチルペンタン

C6H14

96-14-0

1.5

24Methylpentane-2,4-diol, 2-2-メチルペンタン-2,4-ジオール

C6H14O2

107-41-5

4-

Methylpropanoyl chloride, 2-

イソブチリルクロライド

C4H7ClO

79-30-1

6-

Methylpyrrole, N-

1-メチルピロール

C5H7N

96-54-8

0.5

0.8

Methylstyrene

ビニルトルエン

C9H10

25013-15-4

0.5

0.5

Methylthiopropional, 3-3-メチルチオプロピオンアルデヒド

C4H8OS3268-49-3

2-

Mineral oil

ミネラルオイル8042-47-5

0.8

0.7

Mineral spirits

ミネラルスピリット

64475-85-0

0.8

0.7

Monoisobutanolamine

2-アミノ-2-メチル-1-プロパノール

C4H11NO

124-68-5

1.6-

Morpholine

モルホリン

C4H9NO

110-91-822

Myrcene

ミルセン

C10H16

123-35-3

0.5-

Naphtha, hydrotrated heavy

水素化精製重質ナフサ

CnH(2n+2)

64742-48-9

1.0-

Naphthalene

ナフタレン

C10H8

91-20-3

0.4

0.4

Naphthol methyl ether, 2-β-ナフチルメチルエーテル

C11H10O

93-04-9

0.5-

Neopentane

ネオペンタン

C5H12

463-82-1

3.0-

Neopentyl alcohol

2,2-ジメチル-1-プロパノール

C5H12O

75-84-3

2.0-

Nitric oxide

一酸化窒素

NO

10102-43-9

8-

Nitrobenzene

ニトロベンゼンC6H5NO2

98-95-3

1.7-

Nitrogen dioxide

二酸化窒素

NO2

10102-44-0

10-

N-Methylolacrylamide

N-(ヒドロキシメチル)アクリルアミド

C4H7NO2

924-42-5

2.0-

Nonaneノナン

C9H20

111-84-2

1.3

4.7

Nonanol (mixed isomers)

n-ノナノール

C9H20O

143-08-8

1.2-

Nonene (mixed isomers)

ノネン

C9H18

27215-95-8

0.8

-

N

- 115 -

11 付録11-3. VOC 読み替えガスリスト

ガス名（表示名）

ガス名（和名）

分子式

CAS 番号換算
係数(10.6eV)

換算
係数(10.0eV)

Nonene, 1-

1-ノネン

C9H18

124-11-8

0.55-

Norbornadiene, 2,5-

ビシクロヘプタジエン

C7H8

121-46-0

0.6

0.70

Octamethyltrisiloxane

オクタメチルトリシロキサン

C8H24O2Si3

107-51-7

0.3-

Octaneノルマルオクタン

C8H18

111-65-9

1.3

7

Octanol (mixed isomers)

1-オクタノール

C8H18O

111-87-5

1.5-

Octene (mixed isomers)

オクテン

C8H16

25377-83-7

0.9-

Octene, 1-

1-オクテン

C8H16

111-66-0

0.58

1.1

Oxalyl bromide

オキサリルジブロマイド

C2Br2O215219-34-8

5-

Oxydiethanol, 2,2-ジエチレングリコール

C4H10O3

111-46-6

2.0

-

O

P
Paraffin wax, fume

パラフィンワックス

8002-74-2

1-

Paraffins, normal

ノルマルパラフィン

64771-72-8

1-

Paraldehyde

パラアルデヒドC6H12O3

123-63-7

2.0

4.8

Pentacarbonyl iron

鉄カルボニルFeC5O513463-40-6

1-

Pentan-2-one

2-ペンタノン

C5H10O

107-87-9

0.8

1.03

Pentan-3-one

ジエチルケトン

C5H10O

96-22-00.8

0.75

Pentanal

ペンタナールC5H10O

110-62-3

1.2

1.75

Pentandione, 2,4-

2,4-ペンタンジオン

C5H8O2123-54-6

0.8

0.85

Pentaneノルマルペンタン

C5H12

109-66-0

5-Pentanoic acid

吉草酸

C5H10O2

109-52-4

452

Pentanol, 2-2-ペンタノール

C5H12O

6032-29-7

1.5

16

Pentanol, 3-3-ペンタノール

C5H12O

584-02-1

1.5

3.5

Pentene, 1-

1-ペンテン

C5H10

109-67-1

1.3

1.00

Pentylcyclopentan-1-one, 2-2-ペンチル-1-シクロペンタノン

C10H18O

4819-67-4

1-

Pentylcyclopentane

ペンチルシクロペンタン

C10H203741-00-2

1.1-

Pentyne, 1-

1-ペンチン

C5H8

627-19-0

3-

Peracetic acid

過酢酸

C2H4O379-21-0

2-

Perfluorobutadiene

ヘキサフルオロ-1,3-ブタジエン

C4F6

685-63-2

3-

Perfluoro-tert-butylamine

パーフルオロターシャルブチルアミン

C4H2F9N

2809-92-9

5-

Petroleum ether

石油エーテル8032-32-4

0.9-

Phellandrene

パラメンタ-1,5-ジエン

C10H1699-83-2

0.8-

Phenethyl methyl ether, 2-2-フェニルエチルメチルエーテル

C9H12O

3558-60-9

0.6-

Phenol

フェノール

C6H6O

108-95-2

1.2

1.1

Phenoxyethanol, 2-2-フェノキシエタノール

C8H10O2

122-99-6

0.5

10- 116 -

11 付録11-3. VOC 読み替えガスリスト

ガス名（表示名）

ガス名（和名）

分子式

CAS 番号換算
係数(10.6eV)

換算
係数(10.0eV)Phenyl chloroformate

クロロぎ酸フェニル

C7H5ClO2

1885-14-9

1.1-

Phenyl ethyl isobutyrate, 2-イソ酪酸 2-フェニルエチル

C12H16O2

103-48-0

1.5-

Phenyl propene, 2-α-メチルスチレン

C9H1098-83-9

0.4

0.4

Phenyl-2,3-epoxypropyl ether

グリシジルフェニルエーテル

C9H10O2

122-60-1

0.8-

Phenylacetaldehyde

フェニルアセトアルデヒド

C8H8O

122-78-1

0.7-

Phenylacetic acid

フェニル酢酸C8H8O2103-82-2

1-

Phenylcyclohexane

フェニルシクロヘキサン

C12H16827-52-1

0.4-

Phenylethyl acetate, 1-

酢酸 1-フェニルエチル

C10H12O2

93-92-5

0.7-

Phenylethyl alcohol, 2-β-フェニルエチルアルコール

C8H10O

60-12-8

1.2-

Phosphine

ホスフィン

PH3

7803-51-2

2-

Picoline, 3-β-ピコリン

C6H7N

108-99-6

0.9

0.8

Pine oil

パインオイル

8002-09-3

1-

Pinene, α-α-ピネン

C10H1680-56-8

0.27

0.48

Pinene, β-β-ピネン

C10H16

127-91-3

0.27

0.59

Piperazine

無水ピペラジン

C4H10N2

110-85-0

0.8-

Piperidine

ピペリジン

C5H11N

110-89-4

0.9

0.8

Piperylene

1,3-ペンタジエン

C5H8

504-60-9

0.7

1.0

Prop-2-yn-1-ol

2-プロピン-1-オール

C3H4O

107-19-7

2.9-

Propadiene

プロパンジエンC3H4

463-49-0

1-

Propan-1-ol

ノルマルプロピルアルコール

C3H8O

71-23-84.8

40

Propanamide

プロピオン酸アミド

C3H7NO

79-05-02-

Propane-1,2-diol

プロピレングリコール

C3H8O257-55-63-

Propanolamine

3-アミノ-1-プロパノール

C3H9NO

156-87-6

1.5-

Propargyl chloride

プロパルギルクロライド

C3H3Cl624-65-7

2-

Propen-1-imine, 2-2-プロペン-1-イミン

C3H5N

73311-40-72-

Propene

プロピレン

C3H6

115-07-1

1.4

2

Propiolic acid

プロピオル酸C3H2O2471-25-0

8-

Propionaldehyde

プロピオンアルデヒド

C3H6O

123-38-6

1.7-

Propionic acid

プロピオン酸C3H6O2

79-09-48-

Propoxy-2-propanol, 1-

プロピレングリコールモノプロピルエーテル

C6H14O2

1569-01-3

1.1

1.6

Propyl acetate, n-

酢酸ノルマルプロピル

C5H10O2

109-60-4

2.5

17Propyl benzene

1-フェニルプロパン

C9H12

103-65-1

0.5

0.55

Propyl butanoate

酪酸プロピルC7H14O2

105-66-8

2.3

2.7

Propyl formate

ギ酸プロピルC4H8O2110-74-7

10-Propyl iodide

ヨウ化 n-プロピル

C3H7I

107-08-4

1-- 117 -

11 付録11-3. VOC 読み替えガスリスト

ガス名（表示名）

ガス名（和名）

分子式

CAS 番号換算
係数(10.6eV)

換算
係数(10.0eV)

Propylamine, n-

ノルマルプロピルアミン

C3H9N

107-10-8

1-Propylbenzene (all isomers)

プロピルベンゼン（オールアイソマー）

C9H1274296-31-4

0.45

-

Propylene carbonate

プロピレンカーボネート

C4H6O3108-32-7

2-Propylene glycol ethyl ether acetate

プロピレングリコールモノエチル
エーテルアセテート

C7H14O398516-30-4

1.2-

Propylene oxide

酸化プロピレン

C3H6O

75-56-9

2.7-

Propyleneimine

プロピレンイミン

C3H7N

75-55-8

1.3-

Propyneメチルアセチレン

C3H4

74-99-7

4-

Pyrazine

ピラジン

C4H4N2

290-37-9

3-

Pyridine

ピリジン

C5H5N

110-86-1

0.8

0.87

Pyridinol, 4-4-ヒドロキシピリジン

C5H5NO

626-64-2

3-

Pyridylamine, 2-2-アミノピリジン

C5H6N2504-29-0

0.8-

Pyrrole

ピロール

C4H5N

109-97-7

0.6-

Pyrrolidine

ピロリジン

C4H9N

123-75-1

0.4

20

Pyruvaldehyde

メチルグリオキサール

C3H4O2

78-98-80.7-4-メチル-2-（2-メチル-1-プロペニ
C10H18Oル）テトラヒドロピラン

16409-43-1

0.8

-

R
Rose oxide, cisS
Sec-amyl acetate

酢酸 sec-アミル

C7H14O2

626-38-0

2-

Stibine

スチビン

SbH3

7803-52-3

1.5-

Styrene

スチレン

C8H8

100-42-5

0.35

0.52

Terpineol, α-α-テルピネオール

C10H18O

98-55-5

0.8-

Terpinolene

テルピノレン

C10H16586-62-9

0.59

0.9

Terpinyl acetate, α-

酢酸テルピニル

C12H20O2

80-26-2

1.2-

Tert-amyl methyl ether

2-メチル-2-メトキシブタン

C6H14O

994-05-8

0.8-

Tert-butanol

ターシャルブチルアルコール

C4H10O

75-65-0

2.6

2.8

Tert-butyl bromide

2-ブロモ-2-メチルプロパン

C4H9Br507-19-7

1.5

1.6

Tert-butyl formate

ギ酸ターシャルブチル

C5H10O2762-75-4

8-

Tetrabromoethane, 1,1,2,2-

1,1,2,2-テトラブロモエタン

C2H2Br4

79-27-62-

Tetracarbonylnickel

ニッケルカルボニル

NiC4O4

13463-39-3

1-

Tetrachloroethylene

パークロロエチレン

C2Cl4127-18-4

0.44

0.33

Tetrachloropyridine, 2,3,5,6-

2,3,5,6-テトラクロロピリジン

C5HCl4N

2402-79-1

1-

Tetraethyl orthosilicate

テトラエトキシシラン

C8H20O4Si

78-10-4

2

3

Tetrafluoroethylene

四フッ化エチレン

C2F4

116-14-3

15-

T- 118 -

11 付録11-3. VOC 読み替えガスリスト

ガス名（表示名）

ガス名（和名）

分子式

CAS 番号換算
係数(10.6eV)

換算
係数(10.0eV)

Tetrahydrofuran

テトラヒドロフラン

C4H8O

109-99-9

0.8-

Tetrahydronaphthalene

テトラヒドロナフタリン

C10H12

119-64-2

0.4-

Tetrahydropyran

テトラヒドロピラン

C5H10O

142-68-7

3-

Tetrahydrothiophene

テトラヒドロチオフェン

C4H8S

110-01-0

0.6

0.5

Tetramethyl orthosilicate

テトラメトキシシラン

C4H12O4Si

681-84-5

2.0-

Tetramethyl succinonitrile

テトラメチルコハク酸ニトリル

C8H12N23333-52-6

1-

Tetramethylbenzene (all isomers)

1,2,4,5-テトラメチルベンゼン

C10H1495-93-2

0.3-

Tetramethylbutane, 2,2,3,3-

2,2,3,3-テトラメチルブタン

C8H18

594-82-1

1-

Tetramethylgermane

テトラメチルゲルマニウム

C4H12Ge

865-52-1

2-

Tetramethylguanidine,
N,N,N',N'

1,1,3,3-テトラメチルグアニジン

C5H13N380-70-6

0.6-

Tetramethylsilane

テトラメチルシラン

C4H12Si

75-76-3

2-

Thioacetic acid

チオ酢酸

C2H4OS

507-09-5

2-

Thiocarbonyl fluoride

ジフルオロメタンチオン

CSF2

420-32-6

6-

Thiocyanogen

チオシアノーゲン

C2S2N2505-14-6

8-

Thioformaldehyde trimer

1,3,5-トリチアン

C3H6S3291-21-4

1.5-

Thiophene

チオフェン

C4H4S

110-02-1

0.4

0.5

Thiophosgene

チオホスゲン

CSCl2

463-71-8

1-

Thymolチモール

C10H14O

89-83-8

0.7-

Titanium-n-propoxide

テトラプロポキシチタン(IV)C12H28O4Ti

3087-37-4

3-

Toluene

トルエン

C7H8

108-88-3

0.5

0.60

Toluene-2,4-diisocyanate

2,4-トルエンジイソシアネート

C9H6N2O2

584-84-9

1.6-

Toluenesulfonyl chloride, p-

パラトルエンスルホニルクロライド

C7H7SO2Cl

98-59-93-

Toluidine, o-o-トルイジン

C7H9N

95-53-4

0.5-

Tolylaldehyde, p-

パラトルアルデヒド

C8H8O

104-87-0

0.8-

Triazine, 1,3,5-

1,3,5-トリアジン

C3H3N3290-87-9

6-Tributyl phosphate

トリブチルホスフェート

C12H27O4P

126-73-8

5-Tributylamine

トリブチルアミン

C12H27N

102-82-9

1.2

0.6

Trichlorobenzene, 1,2,4-

1,2,4-トリクロロベンゼン

C6H3Cl3120-82-1

0.6

0.5

Trichloroethylene

トリクロロエチレン

C2HCl379-01-6

0.7

0.8

Triethyl phosphate

トリエチルホスフェート

C6H15O4P

78-40-0

3.5-

Triethyl silane

トリエチルシラン

C6H16Si

617-86-7

2-

Triethylamine

トリエチルアミン

C6H15N

121-44-8

0.9

1.1

Triethylbenzene

トリエチルベンゼン

C12H1825340-18-5

0.35

-

Triethylene aluminum

トリエチルアルミニウム

C6H15Al

97-93-8

1-

Trifluoroethene

トリフルオロエチレン

C2HF3

359-11-5

5-- 119 -

11 付録11-3. VOC 読み替えガスリスト

分子式

CAS 番号換算
係数(10.6eV)

2,2,2-トリフルオロエチルメチル
エーテル

C3H5F3O

460-43-5

10-

Trifluoroiodomethane

イオドトリフルオロメタン

CF3I

2314-97-8

2-

Trimethoxymethane

オルソギ酸メチル

C4H10O3

149-73-5

1

10

Trimethoxyvinylsilane

ビニルトリメトキシシラン

C5H12O3Si

2768-02-7

1.0-

Trimethylamine

トリメチルアミン

C3H9N

75-50-3

0.5

0.5

Trimethylbenzene mixtures

トリメチルベンゼン（異性体混合物）

C9H1225551-13-7

0.3

0.3

Trimethylbenzene, 1,3,5-トリメチルベンゼン

C9H12108-67-8

0.4

0.5

Trimethylborate

トリメチルボレート

C3H9BO3

121-43-7

1-

Trimethylcyclohexane, 1,2,4-

1,2,4-トリメチルシクロヘキサ
C9H18ン

2234-75-5

1-

Trimethylene oxide

トリメチレンオキシド

C3H6O

503-30-0

1.5-

Trimethylsilane

トリメチルシラン

C3H10Si

993-07-7

1-

Trioxane

1,3,5-トリオキサン

C3H6O3110-88-3

2-

Turpentine

テレピン油

C10H169005-90-7

0.6

-

TVOC

総揮発性有機化合物11

ガス名（表示名）

ガス名（和名）Trifluoroethyl methyl ether, 2,2,2-換算
係数(10.0eV)U
Undecane

ノルマルウンデカン

C11H241120-21-4

0.9

3.1

Vanillin

バニリン

C8H8O3121-33-5

1-Vinyl acetate

酢酸ビニル

C4H6O2

108-05-4

1.1

1.77

Vinyl bromide

ビニルブロマイド

C2H3Br593-60-2

1.5

0.9

Vinyl chloride

塩化ビニル

C2H3Cl

75-01-4

2.1

1.9

Vinyl ethyl ether

エチルビニルエーテル

C4H8O

109-92-2

0.6

0.95

Vinyl fluoride

ビニルフルオリド

C2H3F

75-02-52-

Vinyl-2-pyrrolidinone, 1-

N-ビニル-2 ピロリドン

C6H9NO

88-12-0

0.9

3.3

Vinylcyclohexene

ビニルシクロヘキセン

C8H12

100-40-3

0.7

0.7

Vinylene carbonate

炭酸ビニレン

C3H2O3872-36-6

1

5Vinylidene difluoride

フッ化ビニリデンモノマー

C2H2F275-38-7

5-Vinylsilane

ビニルシラン

C2H6Si

7291-09-0

1.5-

Xylene mixed isomers

キシレン

C8H10

1330-20-7

0.40

0.59

Xylene, m-

メタキシレン

C8H10

108-38-3

0.4

0.53

Xylene, o-

オルトキシレン

C8H10

95-47-6

0.6

0.6

Xylene, p-

パラキシレン

C8H10

106-42-3

0.4

0.59

Xylidine, all

キシリジン（混合物）

C8H11N

1300-73-8

0.7

0.6

V

X

- 120 -

11 付録11-3. VOC 読み替えガスリスト- 121 -