PC0E-1090

Portable Multi Toxic Gas Detector
MODEL:

SC-9000
3-in-1
Allows up to 3 different toxic gassensors to be installed in a single
unit (3-in-1).
Innovative gas detector■ Simultaneous detection of up to 3 different toxic gases: NH3, HF, Cl2, HCl, O3, SiH4, PH3, etc.
■ Up to 3-year gas sensor warranty: Features next-generation high-performance gas sensors.
■ Outstanding durability: Passes 1.5 m drop testing, equivalent to IP66/68 protection rating■ Bluetooth® equipped! Easy data management via smartphone (optional)■ Inherently safe explosion-proof construction allows use in explosion-proof areas.
■ Continuous operating time: Approx. 60 hours (about twice the time of the previous gas detector)
CE marking compliant

Portable
Multi Toxic Gas Detector
MODEL:

SC-9000

3

Simultaneous Toxic Gas Detection
Features 3 different toxic gas sensors to provide the performance of3 conventional gas detectors in a single unit. Sensor combinations
can be selected from gases including NH3, HF, Cl2, HCl, O3, SiH4,and PH3. Dramatically improves work efficiency while reducingpurchase and maintenance costs.

Up to* For details of detection target gas combinations, please contact RIKEN KEIKI.
Previously...

Each toxic gas typerequired a separate gas detector
NH3 gasdetector

ExampleSC-9000

More compact than previous gas detector

Simply install the required gas sensors
in 3 gas sensor slots!

covers all with asingle unit

NH3 gassensor-22mm

SiH4 gassensor
Cl2 gassensor

SiH4 gas detector

Cl2 gas detector

SC-9000

Previous gas detector

Approx. 154 mm Approx. 132 mm

F sensors installed

3Next-generation high-performance
What are F sensors?

gases

Next-generation high-performance gas sensors that are more compact thanprevious gas sensors while offering greatly improved performance and durability.

Compared to previous
gas sensors, the size
(volume) has been
reduced to less than
1/3.

Previousgassensor

F sensor

Gas sensor
warranty

Up toFeatures next-generation high-performance F sensors offering
outstanding performance and allowing long-term use with peaceof mind.
* Refer to “Gas Sensor Specifications” for details of the specific warranty periods.

yearsBluetooth® equipped!* Easy data management via
smartphone
The RK Link app can be down-

Outstanding durability

1.5m

1.5 m

drop testing
passed2m
1h
Equivalent toIP66/68

protecting rating

50Can communicate with smartphones and tablets via
Bluetooth®. Easy data man℃
agement using the dedicatedOperating temperature range RK Link app to record and-20 °C to +50 °C email measurement results.

〜IP66/68 相当℃-20(Varies depending on gas
sensors installed.)

Snap log button
The snap log
button can beused to save time/
date/user/location/readings.
iOS display example

loaded free of charge from Google
Play or Apple Store.
Bluetooth® and the Bluetooth® logo are
registered trademarks of Bluetooth SIG,
Inc. and are used by RIKEN KEIKI underlicense.

* Please specify when ordering if you require Bluetooth® functionality.

Intrinsically safe explosion-proof construction Continuous operating time: Approx. 60 hours*
Meets Japanese and international explosion-proof standards,
allowing use even in explosion-proof areas.

Twice the time of the previous gas detector to allow use for a week without
recharging, even when used 8 hours a day* Japan Ex, ATEX, IECEx* For lithium ion battery model. Approx. 30 hours for dry cell battery model

Usable in a wide range of fields
With a broad range of toxic gas sensors, the SC-9000 offers outstanding durability andinherently safe explosion-proof construction. You can select target gases dependingon the purpose in various fields such as semiconductor plants, petrochemical plants,shipping, automotive plants, or fumigation.

Semiconductor plant

Petrochemical plant

Shipping

Automotive plant

Google Play and the Google Play logo are trademarks of Google LLC.
Apple and the Apple logo are trademarks of Apple Inc. registered in the U.S. and other countries and regions. App Store is a service mark of Apple Inc. registered in the U.S. and other countries and regions.

Model:SC-9000

Accessories
Tube/strap
Sampling probe

Shoulder strap

Approx. 94 cm tube with gas sampling rod(Interior coating: ETFE)
Allows suction even with strongly
adsorptive toxic gases.

Attaches to the main unit for easy carrying over yourshoulder.
Part No.: 4777 4592 10Part No.: 4733 9406 00

Appearance with accessories attached

Batteries and power related
AC adapterAdapter plug (Type C)

AA alkaline batteries ×6

Part No.: 2594 1342 30

Part No.: 2594 1435 00

Part No. (single battery): 2753 3007 80* Included with rechargeable
battery model* Included with ATEX/IECEx
rechargeable battery model* Included with dry cellbattery model

Optional Accessories
Batteries

Dry battery unit and AA alkaline batteries

Lithium ion battery unit and AC adapter

Inserting batteries allows instant use inemergencies.

The battery unit allows repeated use by recharging.
The AC adapter features a USB Type-C connector.

Dry battery unit

Lithium ion battery unitPart No.:Part No.:(Japan Ex) 4777 9603 60(ATEX/IECEx) 4777 9605 10(Japan Ex) 4777 9602 90(ATEX/IECEx) 4777 9604 30

AA alkaline battery

AC adapter

Part No.: 2753 3007 80

Part No.: 2594 1342 30

Case/holder

Waist belt and waist belt attachment

Sampling rod holder

Aluminum storage case

Allow the gas detector to be worn close to the body.

Attaches to the shoulder strap;
allows the gas sampling rod
tip to be stowed.

Houses the gas detector, accessories,
and optional accessories.

* We recommend using in conjunction
with the shoulder strap to preventthe gas detector dropping.

Dimensions:
Approx. 365 mm (W) × 236 mm (H) × 226 mm (D)*Part No.: 4775 5651 00

Waist belt

Part No.: 4777 9579 00

Part No.: 4775 5653 40* excluding projections

Waist belt attachment
Part No.: 4775 9853 10
Waist belt
attachment

Waist belt

Management software/cable

USB cable (1 m)

Data logger management program

Connects the gas detector to a PC.
Used when using the software

Software used to view and manage
measurement results and logs of events suchas alarms and calibrations

Part No.: 2440 2728 90

Part No.:

Simply install thesoftware on a PC.

(Japan Ex) 9811 0980 90(ATEX/IECEx) 9811 0990 80

Configuration Program
A configuration program to edit SC-9000
settings can be downloaded free ofcharge from the RIKEN KEIKI website.

Example: Measurement Example: Measurement
results (table)
results (graph)

Miscellaneous

Adapter plugs

Protective film

The Type A AC adapter can beconverted to Type C, O, or BF.

To protect the LCD(set of 5)

Part No.:Part No.: 4777 9025 70(EU/Type C) 2594 1435 00(AU/Type O) 2594 1434 20(UK/Type BF) 2594 1436 70

Product Specifications
Specifications

Item

LCD digital (full dot)

Concentration display

Up to 3 different gas sensors can be installed.
(NH3, HF, Cl2, HCl, O3, SiH4, PH3, NO, NO2)

Detection target gases*1

Detection method

Pump suction type

Suction flow rate

Minimum 0.75 L/min (open flow rate)
Clock, battery level, operation status

Display items

English, Cantonese (Traditional Chinese), Czech, French, German, Italian, Japanese, Korean, Mandarin (Simplified Chinese),
Polish, Portuguese, Russian, Slovak, Spanish, Turkish, Vietnamese

Display languages

Approx. 95 dB (mean value at 30 cm from source)

Buzzer volume

Lamp flashing, continuous modulating buzzer sounding, gas concentration reading blinking

Gas alarm indication

Self-latching, auto-reset

Gas alarm pattern

Flow rate abnormality, system abnormality, sensor abnormality, low battery voltage, calibration failure, clock abnormality

Fault alarm/self-diagnosis
Fault alarm indication

Lamp flashing, intermittent buzzer sounding, detail display

Fault alarm pattern

Self-latchingUSB 2.0 Type-C (for data logger/setting), Bluetooth 4.2 (Bluetooth Low Energy)

Communication specifications

Dedicated lithium ion battery unit (BUL-9000) or dedicated dry battery unit (AA alkaline batteries ×6) (BUD-9000)

Power source*2

Lithium ion battery unit: Approx. 60 hours (at 25 °C, no alarm, no lighting)
Dry battery unit: Approx. 30 hours (at 25 °C, no alarm, no lighting)Continuous operating time

-20 °C to +50 °C (no sudden fluctuations)Operating temperature range*3
Operating humidity range*3

20 to 90 %RH (no condensation)

Operating pressure range

80 kPa to 120 kPa (80 kPa to 110 kPa for explosion-proof range)
Dustproof and waterproof construction equivalent to IP66/68, drop resistant to 1.5 m

Construction*4

Intrinsically safe explosion-proof construction

Explosion-proof construction

Japan Ex:ATEX:
IECEx:

Explosion-proof class*5

CE marking

Certification

Approx. 158 mm (W) × 85 mm (H) × 132 mm (D) (excluding projections)

External dimensions

Approx. 1.1 kg

Weight*6
*1:
*2:
*3:
*4:
*5:
*6:

Ex ia IIC T4 Ga
II 1 G Ex ia IIC T4 Ga
Ex ia IIC T4 Ga

For details of detection target gas combinations, please contact RIKEN KEIKI.
To ensure explosion-proof performance, use only the batteries indicated on the explosion-proof certificate.
May vary depending on the gas sensors installed.
IPx8: No water penetration when submerged at depth of 2 m for 1 hour.
For dry cell battery model, the explosion-proof class may differ in some cases when using battery power.
Including batteries and battery unit.

Gas Sensor Specifications
Detection
targetgas
NH3

0 – 75.0 ppm

0.5 ppm

25.0/50.0 ppm

2 years

Detection
target
gas
SiH4

HF

0 – 1.50 ppm

0.01 ppm

0.50/1.00 ppm

1 yearPH3

Detection range

Resolution

Alarm setpoints(1st/2nd)

Warranty
periodDetection range

Resolution

Alarm setpoints(1st/2nd)

Warranty
period0 – 15.0 ppm

0.1 ppm

5.0/10.0 ppm

3 years0 – 1.00 ppm

0.01 ppm

0.30/0.60 ppm

3 years

Cl20 – 1.50 ppm

0.01 ppm

0.50/1.00 ppm

3 years

NO0 – 100 ppm

1 ppm

25/50 ppm

3 years

HCl0 – 6.00 ppm

0.05 ppm

2.00/4.00 ppm

3 years

NO20 – 15.0 ppm

0.1 ppm

5.0/10.0 ppm

3 years

O30 – 0.600 ppm0.005 ppm

0.100/0.200 ppm

1 yearDistributed by:
2-7-6 Azusawa, Itabashi-ku, Tokyo 174-8744, Japan

Phone : +81-3-3966-1113
Telefax : +81-3-3558-9110
E-mail : intdept@rikenkeiki.co.jp
Web site : https://www.rikenkeiki.co.jp/english※ The contents described in this catalog are subject tochange without notice according to the performance
improvement.

PC0E-1090-240600