Portable Hydraulic Pressure Calibrator

PCS-H100

Portable Pressure Calibrator

Vent/release valve

Test port

Electrical connections formeasuring mA, V anddetecting switch contacts.

On/OffHand strap

Full color touch
screen display

Lever type hand pump

Fine pressureadjustment

Oil Reserver
Pressure range

PCS-H100 is Portable pressure calibrator

for Hydraulic pressure calibration up to 1000 bar
with pressure generation and precise control.
An independent pressure calibration system
combining electrical signal measuring and loop
power to quickly and easily calibrate for a large· 0.02% F.S Accuracy· Electrical connection for Transmitter-Measure up to ±15 V, ±24 mA-Loop Power· Pressure switch test, Help functions

Using lever type hand pump to generate up to· Hydraulic Quick adaptor set (Optional)

the built-in volume controller to precisely adjust
the pressure with the secondary fine pressurecontrol.
Full-graphic touchscreen display with intuitive
menu selection makes it high visibility andoperability.
Upper side of PCS-H100 can equipped quickconnectors and adaptors for quick and easy toconnect DUT.

Solution for Pressure Measurement & Calibration

Easy calibration task settings (Touch Screen)-Power 24 VDC

amount of on-site calibration.

1000 bar and quickly and easily, and then use

35· Lever type hand pump to generateand control up to 1000 bar Hydraulic· Included carrying case
· CE, ROHS CertificatedUse· Portable pressure calibration· Simple calibration for common calibration lab· Calibrate pressure gauge for pressure measuring
instruments retail business· Pressure generating and controlling for pressure
test lab· Pressure calibration for laboratory

Main Screen

Pressure/ElectricalMeasurement Screen

Help ScreenApplication connection
diagram display function

Pressure Units Selection
Available to select13ea of pressure units· Hydraulic high pressure calibration· Easy & quick calibration for pressure transmitter,
analog pressure gauge, digital pressure gauge,pressure switch test
Solution for Pressure Measurement & Calibration

36

Portable Hydraulic Pressure Calibrator

PCS-H100

Portable Pressure Calibrator

TASK menu (Advanced features)

Ordering Information
Model(PCS-H100_Range) / Description(Portable Hydraulic Pressure Calibrator / Media)

20 MPa0 MPa to 20 MPa

35 MPa0 MPa to 35 MPa

70 MPa0 MPa to 70 MPa

100 MPa

0 MPa to 100 MPa

W(Water)
O(Oil)

Accessories
Task Menu

Min/Max Test

Leak Test

Switch Test

‘ Assembling
External
Pressure
Manifold ’General Specifications
Pressure range
Accuracy
Electronic measurement
Sensor supply power
Display
Power±0.02% F.S±15 V.dc, ±24 mA.dc, Accuracy ±(0.01% of Reading + 1digit), Loop Power

Resolution ±999999 (Available to select 4 to 6 digits.)
Lithium-ion rechargeable battery, Charger 9V, 6A (use 30 hours on a 5 hours charge)

Storage
Temperature Range-30 to 80 °C (5 to 75 °C with water)

TemperatureCompensated Range

0 to 50 °C (5 to 75 °C with water)bar, mbar, Pa, hPa, kPa, MPa, kg/cm2 , psi, mmH20, cmH20, inH20, mmHg, inHg, mSW, fSW(mSW and fSW units are applied on a water temperature of 15°C with reference to U.S. Navy
Dividing Manual, Revision 7, Table 2-10. Pressure Equivalent.)

Test Port

1/4" PF Female

Weight4.2 kg including batteries

Oil Reserver Volume
Media

Rechargeable battery

Battery charger

Test lead set

Full color touch screen LCD, 110 mm(4.3inch) diagonal. 480 x 272 pixels.

0 to 70 °C (5 to 75 °C with water)

Size

Carrying casePower 24 VDC

OperatingTemperature Range

Pressure units

37

0 to 200 bar...1000 bar

Optional items

Hydraulic Quick adaptor set(up to 2000 bar)

Flexible tubes(up to 630 bar)

Traceable calibration
report with data (KOLAS)

CA-H Series( 1/8, 1/4, 3/8, 1/2 PT | PF, NPT)FT Series(30 cm to 300 cm)

Pressure

External Pressure
Manifold

210 mm(W) × 360 mm(D) × 110 mm(H)100 ccWater or Mineral Oil

Solution for Pressure Measurement & Calibration

Solution for Pressure Measurement & Calibration

38