MODEL 8011-01 & 8012-01
ACCELEROMETERSPECIFICATIONSdimensions

IEPE Accelerometer
16kHz Bandwidth-55°C to +125°C Operating Range
Top Exit Connector or Integral Cable

The Model 8011-01 & 8012-01 are internallyshielded rugged IEPE accelerometers designed for harshenvironments. The accelerometers are available in ±10g,±80g & ±500g dynamic ranges and have a flat bandwidth up
to greater than 16kHz. The model 8011-01 features a top
exit MIL-C-5015 connector while the model 8012-01
features an integral cable. The units are hermetically sealed
and are designed to operating in temperature ranges from 55°C to +125°C.
FEATURES

±10g, ±80g & ±500g Dynamic Ranges
Wide bandwidth up to 16kHz
Case Isolated, Internally Shielded
Hermetically Sealed, Welded
Annular Shear Mode
Reverse Wiring Protection
Stable Temperature Response

APPLICATIONS





SENSOR SOLUTIONS /// Model 801X-01 Rev B

General Purpose
Machine Monitoring
Industrial Applications
Harsh Environments
Gearbox Monitoring

9/2015

Page 1

MODEL 8011-01 & 8012-01 ACCELEROMETERPERFORMANCE SPECIFICATIONS
All values are typical at +24°C, 80Hz and 4mA excitation unless otherwise stated. Measurement Specialties reserves the right to update
and change these specifications without notice.
Parameters
DYNAMIC
Range (g)Sensitivity (mV/g)
Frequency Response (Hz)Frequency Response (Hz) 2
Natural Frequency (Hz)Non-Linearity (%FSO)
Transverse Sensitivity (%)
Shock Limit (g)±105001-40000.5-7000
28000±1<55000ELECTRICAL
Compliance Voltage (Vdc)
Excitation Current (mA)
Bias Voltage (Vdc)
Output Impedance (Ω)
Insulation Resistance (MΩ)
Residual Noise (g RMS)
Warm-up Time (sec)Grounding

18 to 30
18 to 3018 to 30
2 to 102 to 102 to 108 to 128 to 128 to 12<100<100<100>100>100>1000.00005
0.00010.0003<1<1<1Case Isolated, Internally Shielded

ENVIRONMENTAL
Temperature Response (%)
Operating Temperature (°C)
Storage Temperature (°C)

±5 (see typical temperature response on following page)-55 to +125
-55 to +125

PHYSICAL
Sensing Element
Case Material
Weight (grams)
Mounting TorqueCeramic (shear mode)
Stainless Steel93
24 lb-in (2.7 N-m)

Notes
±80
1001-10000
0.5-16000
32000±1<55000±500
101-100000.5-18000
42000±1<550001

For full dynamic range a minimum compliance voltage of 24Vdc is recommended

2

LF option offers a low frequency response to 0.1Hz

3Contact factory for Hazardous Area Approval options±10% (±5% Option)±10%±3dB

See Note 1
Room Temperature@100Vdc

Calibration supplied:

CS-FREQ-0100

NIST Traceable Amplitude Calibration from 20Hz to ±10% Frequency Response Limit

Supplied accessories:

AC-A03663¼-28 to ¼-28 mounting stud

Optional accessories:

316A-XXX
318A-XXX
AC-D03664
AC-D03665

Cable Assembly, Straight (XXX designates length in inches, 10ft standard)
Cable Assembly, 90 Degrees (XXX designates length in inches, 10ft standard)¼-28 to M5 mounting stud¼-28 to M6 mounting stud

The information in this sheet has been carefully reviewed and is believed to be accurate; however, no responsibility is assumed for inaccuracies. Furthermore, thisinformation does not convey to the purchaser of such devices any license under the patent rights to the manufacturer. Measurement Specialties, Inc. reserves theright to make changes without further notice to any product herein. Measurement Specialties, Inc. makes no warranty, representation or guarantee regarding thesuitability of its product for any particular purpose, nor does Measurement Specialties, Inc. assume any liability arising out of the application or use of any product or
circuit and specifically disclaims any and all liability, including without limitation consequential or incidental damages. Typical parameters can and do vary in different
applications. All operating parameters must be validated for each customer application by customer’s technical experts. Measurement Specialties, Inc. does notconvey any license under its patent rights nor the rights of others.

SENSOR SOLUTIONS /// Model 801X-01 Rev B

9/2015

Page 2

MODEL 8011-01 & 8012-01 ACCELEROMETER

Typical Frequency Response
4035
30
25

151050-5-10-15-20-25-30

Standard Option-35

LF Option

-40
0

1

10100

1000

10000

100000Frequency (Hz)Typical Temperature Response
10%8%6%

Sensitivity Deviation (%)

Sensitivity Deviation (%)

20

4%2%0%-2%-4%-6%-8%-10%-60-40-200204060

80

100120140

Temperature (°C)

SENSOR SOLUTIONS /// Model 801X-01 Rev B

9/2015

Page 3

MODEL 8011-01 & 8012-01 ACCELEROMETER

ORDERING INFORMATIONPART NUMBERING

Model Number+Range

801XLF-01-GGG
l ll_____ Range (080 is 80g)
l l__________ Low frequency option, otherwise leave blankl___________ Electrical Connection Option (1 for 2-pin connector, 2 for integral cable)
Example: 8011-01-080
Model 8011-01, 2-pin connector, 80g range

STANDARD CONFIGURATIONSPart NumberElectrical ConnectionRange

Cable Length

8011LF-01-010

Connector

10g

Not Applicable8011-01-080Connector

80g

Not Applicable8011-01-500

Connector

500g

Not Applicable

8012LF-01-010-120

Integral Cable

10g

10 Feet8012LF-01-010-240

Integral Cable

10g

20 Feet8012LF-01-010-360

Integral Cable

10g

30 Feet8012-01-080-120

Integral Cable

80g

10 Feet8012-01-080-240

Integral Cable

80g

20 Feet8012-01-080-360

Integral Cable

80g

30 Feet8012-01-500-120

Integral Cable

500g

10 Feet8012-01-500-240

Integral Cable

500g

20 Feet8012-01-500-360

Integral Cable

500g

30 FeetTE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred to herein
might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard
Terms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuseof the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies

All Rights Reserved.

SENSOR SOLUTIONS /// Model 801X-01 Rev B

9/2015

Page 4