®
R&S RT-Z2T

Probe Interface Adapter
Manual(=J[á2)Version 02

Manual1326438302

This manual describes the R&S RT-Z2T probe interface adapter (1338.0007.02).

© 2018 Rohde & Schwarz GmbH & Co. KG
Mühldorfstr. 15, 81671 München, Germany
Phone: +49 89 41 29 - 0
Fax: +49 89 41 29 12 164
Email: info@rohde-schwarz.com
Internet: www.rohde-schwarz.com
Subject to change – Data without tolerance limits is not binding.
R&S® is a registered trademark of Rohde & Schwarz GmbH & Co. KG.
TEKTRONIX, TEK, and TEKPROBE are registered trademarks of Tektronix, Inc.
Trade names are trademarks of the owners.
Throughout this manual, products from Rohde & Schwarz are indicated without the ® symbol, e.g.
R&S®RT-Z2T is indicated as R&S RT-Z2T.

R&S®RT-Z2T

Product Description
Usage Information

1Product Description

The R&S RT-Z2T probe interface adapter is a product designed to connect a series of active Tektronix oscilloscope probes with TekProbe BNCTM level II interface
to every R&S oscilloscope that is equipped with the Rohde & Schwarz probeinterface.
The adapter provides power to operate the probe. The R&S oscilloscope is adjusted to the correct settings by selecting the probe from a list of predefined probes
in the probe menu of the oscilloscope.

1.1

Supported Tektronix Probes

The following probes are supported by the R&S RT-Z2T adapter:● Single-ended probes: P6205, P6243, P6245, P6241, P6249● Current probes: TCP202● Differential probes: P6246, P6247, P6248, P6250, P6251● High-voltage differential probes: P5205, P5210● Electro-optical probes: P6701B, P6703B, P6711, P6713

1.2

Usage InformationThe R&S RT-Z2T probe adapter is an add-on probe adapter, indented to be usedat laboratory or test and measurement areas. These areas are used for analysis,testing and servicing and where equipment is operated by trained personnel.

Manual 1326.4383.02 ─ 02

3

R&S®RT-Z2T

Product Description
Inspecting the Contents

Risk of injury
Use the accessory carefully to prevent electric shock, personal injury, ordamage. Read and observe the "Basic Safety Instructions" that are delivered as a printed brochure.
Also, read and observe the safety instructions of the oscilloscope theaccessory is connected to.

1.3

Deliveries

The R&S RT-Z2T delivery package contains the following items:● R&S RT-Z2T probe interface adapter● Manual● "Basic Safety Instructions" brochure

1.4

Inspecting the Contents● Inspect the package for damage.
Keep a damaged package and the cushioning material until the contents havebeen checked for completeness and the device has been tested.
If the packaging material shows any signs of stress, notify the carrier and yourRohde & Schwarz service center. Keep the package and cushioning materialfor inspection.
● Inspect the product.
If there is any damage or defect, or if the R&S RT-Z2T does not operate properly, notify your Rohde & Schwarz service center.

Manual 1326.4383.02 ─ 02

4

R&S®RT-Z2T

2Device Description

Device Description

The R&S probe interface of the R&S RT-Z2T probe interface adapter is attached
to a Rohde & Schwarz oscilloscope.

Figure 2-1: R&S probe interface of the R&S RT-Z2T
1 = Release knob
2 = Rohde & Schwarz probe interface with 7 mm (276 mil) coaxial connector and 6 pogo pins

Risk of interface damage
Connect the adapter only to a base unit that has a Rohde & Schwarz probe
interface. Do not connect the adapter to a commercially available BNC
jack, as this connection can damage the interface irreparably. From the outside, the 7 mm coaxial connector of the R&S probe interface looks like astandard BNC connector. However, it is constructed differently and has asignificantly higher frequency limit.
The Tektronix active probe is connected to the adapter’s TekProbe BNCTM level IIinterface. This interface also provides the power for the probe.

Manual 1326.4383.02 ─ 02

5R&S®RT-Z2T

Putting into Operation

Figure 2-2: TekProbe BNCTM level II interface of the R&S RT-Z2T

3Putting into Operation

The R&S RT-Z2T has been designed to withstand a moderate amount of physicaland electrical stress. Treat the adapter with care.

Risk of adapter and instrument damage due to excess power
Voltages above the specified limits of the R&S RT-Z2T probe interface
adapter or the connected R&S oscilloscope can damage the adapter or theoscilloscope. Do not exceed the lesser of the maximum input voltages
specified for the adapter or the oscilloscope.
If the oscilloscope is set to 50 Ω, input coupling, the maximum input voltage
is usually determined by the oscilloscope. If the oscilloscope is set to 1 MΩ
input coupling, the maximum input voltage is usually determined by theadapter.

Manual 1326.4383.02 ─ 02

6

R&S®RT-Z2T

Putting into Operation
Connecting the Adapter to an R&S Oscilloscope

Damage caused by electrostatic discharge
To avoid damage of electronic components, protect the operational siteagainst electrostatic discharge (ESD).

Wristwithwith
cord cord
Wriststrap
strap

Building ground

FloorFloor mat
mat

Heelstrap
strap
HeelGroundconnection
Groundconnection
ofsite site
ofoperational
operationalTo protect the operational site, use one or both of the following measures:● Wrist strap with cord to ground connection● Conductive floor mat and heel strap combination

3.1

Connecting the Adapter to an R&S Oscilloscope

The R&S RT-Z2T has been designed for use with R&S oscilloscopes that feature
an R&S probe interface.

Manual 1326.4383.02 ─ 02

7

R&S®RT-Z2T

Putting into Operation
Connecting the Tektronix Probe to the Adapter► Connect the adapter (1) to the Rohde & Schwarz probe interface of the baseunit (2). The adapter snaps in when connected properly to the port.
► To disconnect, press the release button (3). Pull the adapter away from thefront panel of the base unit.

3.2

Connecting the Tektronix Probe to the Adapter

To connect a Tektronix probe to the probe interface of the R&S RT-Z2T adapter:
1. Turn the lever of the probe to the left side in the open position.
2. Connect the BNC connector of the probe to the BNC socket of the R&S RTZ2T adapter.
3. Lock the probe by pushing the lever to the right into the closed position.
4. Adjust the oscilloscope settings by choosing the correct predefined probe orby entering the values for external attenuation, input coupling and display unit
manually.
For a description of the usage of a Tektronix probe, refer to its user manual.

Manual 1326.4383.02 ─ 02

8

R&S®RT-Z2T

Maintenance and Service
Returning for Servicing

4

Maintenance and Service

4.1

Service Strategy

Like all Rohde & Schwarz products, Rohde & Schwarz probes abd adapters areof high quality and require only minimum service and repair. However, if service isneeded, contact your Rohde & Schwarz service center. Return a defective product to the Rohde & Schwarz service center for diagnosis and exchange.
You can return the R&S RT-Z2T probe interface adapter for calibration. The service personnel will perform the required tests.

4.2

Returning for Servicing

Use the original packaging to return your R&S RT-Z2T to your Rohde & Schwarz
service center. A list of all service centers is available on:www.services.rohde-schwarz.com
If you cannot use the original packaging, consider the following:
1. Use a sufficiently sized box.
2. Protect the product from damage and moisture (e.g. with bubble wrap).
3. Use some kind of protective material (e.g. crumpled newspaper) to stabilize
the product inside the box.
4. Seal the box with tape.
5. Address the package to your nearest Rohde & Schwarz service center.

Manual 1326.4383.02 ─ 02

9

R&S®RT-Z2T

Maintenance and Service
Discarding the Product4.3

Cleaning

Product damage caused by cleaning agents
Cleaning agents contain substances that may damage the product; forexample, solvent may damage the labeling or plastic parts.
Never use cleaning agents such as solvents (thinners, acetone, etc.), acids,bases or other substances.

Product damage caused by static electricity
Before cleaning the facing surfaces of the sensor head, discharge any static
electricity at your hands. Thus you ensure that no high voltage caused bystatic electricity is applied to the product. Application of a high voltage may
damage the internal Hall elements or circuitry. You can attract static electricity to your hands by touching a nearby metal object.
To clean the exterior of the probe or adapter, use a soft cloth moistened witheither distilled water or isopropyl alcohol. Before using it again, make sure to dry it
completely.

4.4

Discarding the ProductHandle and dispose the product in accordance with local regulations.

Manual 1326.4383.02 ─ 0210