LE-3500
Multi Protocol Analyzer
LE-2500
Battery-powered portable communications
analyzer in a newly designed compact frame.

MULTI PROTOCOL ANALYZER

LE-3500
● 210
（W）
×154
（D）
×38
（H）
mm,
Approx. 790 g

ModbusSupported

MULTI PROTOCOL ANALYZER

LE-2500
● 210
（W）
×154
（D）
×38
（H）
mm,
Approx. 760 g

Standard
communications ＞＞＞
Expanded
communications

RS-232C

RS-422/485 (RS-530)

C/S PI IrDA/ASK Current Loop＞＞＞ TTL/I2C/SPI
CC-Link

CAN/LIN

Battery-powered Portable Communications Analyzer
RoHS Directive
Compatible

MULTI PROTOCOL ANALYZER

LE-3500 / LE-2500

A product incorporating features
for the protection of the global
environment, such as theelimination of toxic substances,
and a power-saving design.

LINEEYE released the first LE-series model in 1986. Since then, LINEEYE has been developing the LE Series in responseto the requirements of a large number of customers, and now come to the fruition of the LE-3500 and LE-2500, the newly
designed latest models realizing great downsizing with a sharp reduction in weight. The LE-3500 and LE-2500 incorporate
versatile analysis functions and excellent portability, thus fully supporting the trouble analysis of communications systems,
industrial equipment, and a variety of in-vehicle networks as well as development tests and after-sale services.
External signal I/O terminal
RS-422/485 (RS-530) measurement port
RS-232C (V.24) measurement port
DC input connector (polarity insensitive)

Exchangeable measurement board

AUX (RS-232C) connector
Elastomer resin

Power supply
switchUSB2.0 connector

Contrast
adjustment
volumeCF card slot

Special data items are expressed in individual signs.

Menu keyLine state LED(Example of display signs)
HDLC/SDLC flag
Short frame

Measurement
start key

Idle time (frame interval time)

Block check OK
Block check NG
Parity error

Time stamp (frame arrival time)Continuously displays the state ofcommunications line with a 2-color LED.

In RS-232C
Red: ON space state
communications
Green: OFF mark state
OFF: Not in use or not wired

Framing error
PE and FE simultaneous error
BreakSPEED In half-duplex mode
Max speed
4Mbps

4M bps100MB
Memory2M bps
1 M bps

Max speed
1Mbps

Max speed
500Kbps

500Kbps

8GB
CF card support

8GB
CF card support

LE-1500
Design for Async communication equippedwith sufficient measuring functions.

6.4MB
MemoryUSB Flash Drive(LE-8200A)

MULTI PROTOCOL ANALYZER

LE-3500
MULTI PROTOCOL ANALYZER

COMPACT PROTOCOL ANALYZER

64GB
CF card support

16GB
CF card support

2.4MB
Memory2.4MB
MemoryMax speed
2Mbps

LE-2500

A perfect model incorporating statisticalanalysis and program simulation functions.

MULTI PROTOCOL ANALYZER

LE-8200A
LE-8200
The high-end model boasting plentiful performancewith a large-sized color display and versatilefunctions.
*See the LE-8200 catalog for details.

A high cost-performance multi-protocol
model with expandability.

*See the LE-1500 catalog for details.

Async only（Async, PPP）

Multi Protocol（Async, Sync, BSC, SDLC,HDLC, X25, PPP, I2C, IrDA, CAN, LIN ････）

A measurement tool inevitable to on-site tests as well as analysis ofcommunications line trouble.

A lightweight unit in A5 size that can bebattery driven for 8 hours continuously.
●

Great downsizing w
ith a●n weight.
tion i
duc
p re

sha
r

Conventionalmodel

NEW LE-3500

LE-3200
▲
▲
▲
▲
▲
▲

B5 size
240(W)×180(D)×39(H)mm

A5 size
210(W)×154(D)×38(H)mm

Multi-protocol support

Mega Speed Measurement

Incorporates RS-232C and [Protocol setting display]
RS-422/485
measurement
interfaces
asstandardfeatures and supports avariety of communications
protocols. A DSUB 25-pin
terminal block and a variety of[Interface setting display]
dedicated cables, such as
X.25, RS-449, and V.35 (*)
cables, are available.

Analysis is possible at any [Baud rate setting display]baud rate(*) from low speed
to high speed. Margin tests oncommunication speed deviation
are simple.
* Using high precision DPLL technology
for open baud rate support, transmission
and reception speeds can beseparately set to an effective 4 digits.

* LE-2500 does not support Modbus andV.35 control signals.

Auto Save/Long Recording Time

Expansion Kits support TTL, I2C, SPI, IrDA, CAN, and LIN
Protocols of differing hardware specifications are supported by simply
changing the measurement boards.

TTL, I2C, SPI

IrDA, ASK

CAN, LIN

Expansion kit OP-SB5GL

Expansion kit OP-SB6G

Expansion kit OP-SB7GX

You can record communications data endlessly or stop recording it automatically when thememory is full. Furthermore, an auto save function makes it possible to save the monitoredcontent of captured memory on a CF card. Auto Save continuously saves data into themeasurement log of a user-specified file size, using ring recording as long as the card has
space. It is useful for identifying rare communication failures of unknown cause.
Target line Continuous recording time reference（LE-3500）
speed (bps) Main memory only When using CF-16GX

andSignalVoltageCommunication line timing is analyzed and displayed as a logic analyzer display to a time
resolution of max. 50 ns. The new function of signal voltage measurement ensures ease of thevoltage measurement of RS-232C signals in places where tester probes cannot reach smoothly.
[Example of logic analyzer measurement]

Approx. 22 min

Approx. 960 hrs

Approx. 110 sec

Approx. 80 hrs

1MApprox. 14 sec

Approx. 10 hrs*Calculated for full-duplex transmission of 1,000 byte data
frames per second. Both transmission and reception data
consume 4 byte of memory with each capture.

>>> See pages 8 and 9 for details.

Logic Analyzer
Measurement

9600
115.2K[Example of signal voltage measurement]

PC Link
Text conversion software and capturing software for printout
data (*) , makes it possible to utilize measurement data on
the user's PC. Furthermore, the use of the optional PC
Link Software will widen the application range.

>>> See pages 7 for details*: These software can be downloaded from LINEEYE's website for free.

A monitor function to visualize communications data.
Supports multi-protocols
The online monitor featurerecords communications data inthe capture memory and provides
an easy-to-understand display
for the type of protocol, without
affecting the communications
line. As a standard feature,
LINEEYE protocol analyzers
support various communications
standards from asynchronoustopacketcommunication.
Depending on the test, you canselect bit transfer sequence andpolarity, as well as modulation
format from NRZ, NRZI, FMO,
FM1, and 4PPM. The featureallows to support effective
analysis by omitting SYN codes
and using SDLC/HDLC addressfilter.

Trigger Feature for Catching User-specified Events
The trigger feature allows you
to specify a communications
event as the trigger conditionand have measurement operations
executed automatically whenthat condition is satisfied. Up tofour pairs of conditions andoperations can be set, which is
helpful towards identifying
frequent intermittent faults that
occurs with communications
systems. And, the operation ofa trigger condition can be
specified as the condition foranother trigger, making it
possible to analyze complicatedoperations based on sequential
triggers.

[Data display][Example X.25 protocol translation][Example PPP translation][Trigger setup summary display][Example trigger condition setup][Example trigger action setup]

Line state LED[Example of connection for online monitoring]monitor cableCommunications
network
DCE

RD

Beforeconnection

Communications line state is indicatedin real time using 2-color LEDs

DTE

SD

SD RD

After
monitorcable
coccection

LINEEYE protocol analyzers
can analyze communications
data and automatically set
basic measuring conditions,such as communications speed,
character framing, data code,
synchronization
character,
BCC/FCS, etc. This is effective
for monitoring lines of unknown
communications conditions.

[Monitor condition auto setting – Search display][Monitor condition auto setting – Determination display]*The auto setting is not accurate withsmall volumes of communications
data or data that contains manyerrors.

Records Time Data with Communication Data
LINEEYE protocol analyzers
record not only communications
data but the time (time stamp)of transmissions and receptions
as well as idle time; therefore
failure time and timeout status
can be checked. It is also
possible to record the informationof changes in control lines atthe same time. For ASYNC/
BURST communications, setting
the idle time to be the frame end
is available in the rage of 1 to
100ms.

Monitor Condition Auto Setting

Delay time function added with a voltage measurement feature[Example record data selection]

A feature to measure the voltage
of four RS-232C signal lines has
been added to the conventional
delay time function used toanalyze the delay time of control
line changes (e.g., RTS to CTSchanges) at a resolution of 0.1 ms.

[Time stamp / idle time display][Example delay time measurement ]Statistical Analysis Capabilities
Statistics can be compiled fortransmission and reception data
sets, frames and the number of
established trigger events, andsubsequently displayed as a graph(Unit: 1-240 min.). This helps tounderstand communications
traffic and error frequency fora specific time period.

[Timing data display]

Only for LE-3500[Graph of statistically analyzed data]

BERT function to measure the occurrence rate of communications errors.
BERT function enables you tomeasure transmission quality ofcommunications lines by aloop-back or interactive connection.
It is possible to measure evaluation
parameters (bit error count,block error count) conforming to
ITU-T G.821 Notification, hence
enabling bit error rate evaluations
and fault point identification.
Elaborate test patterns andfunctions such as bit error forced
interrupt are comparable todedicated equipment.

Evaluation is possible in ASYNC
or SYNC mode, by specifying
measurement period (continuous,received bits, specified time,
repeat) or test pattern.

[Example connection for BERT]
Loop-back test

Test
patternCommunications
network

Interactive test

Test
patternCommunications
network

Testpattern

Loopback
point

Once started, the results ofmeasured line quality are displayed
and updated in real time. Whenfinding the error bit, LINEEYE
analyzers can output the external
trigger to inform other equipments.
Repeat mode allows you to know
the error rate for the specific time
range in the communication lines.
[Contents of BERT measurement]
Savail Available measurement in seconds[BERT setting display][Example BERT measurement]

0〜9999999

Loss

SYNC loss count

R-Bit

Effective bits received

0〜9999999〜9.99E9

R-Blk

Effective blocks received 0〜9999999〜9.99E9

E-Bit

Error bit countBlock error count0〜99990〜9999999〜9.99E9

E-Blk

Bit-ER Bit error rate

0〜9.99E-9〜1

Blk-ER Block error rate

E-Sec Error in seconds

0〜9999

%E.F.S Normal operation rate 0.000〜100.000%

0〜9999999〜9.99E9
0〜9.99E-9〜1

Example of connecting
RS-485 signals via optionalterminal block (LE-25TB) fora DSUB 25-pin connector.

LE-25TBMULTI PROTOCOL ANALYZER

LE-3500 / LE-2500

Online Monitoring,
Simulation and
BERT Support All-in-One
Simulation function to conduct transmission and reception tests in place of target equipment under test.
With the simulation feature, the LINEEYE protocol analyzers act as the counterpart tothe target device and perform transmission and reception tests according to protocol.
Even in the early stages of development when matching devices are not available,
tests can be run at near to actual operating status. After checking the communications
protocol step by step in LINEEYE[Control lines and data transmission timing]
analyzer’s own original MANUAL
mode, a developer can create a
simple program to branch
conditions using menu selection
and test more complicatedcommunications
protocols.
Communications speed can be[Example of connection for simulation]
freely set ; therefore margins can beevaluated by intentionally shifting
communications speed, and error
Test data
response processing can bechecked using test data thatmixes in data with parity errors.
In addition, data transmission
can be linked with the changes
in the signal lines such as RTS
and CTS at the preset timing.

■ POLLING mode
The POLLING mode simulates the slave and master units in multidrop (1:Nconnection) polling protocols. In the slave mode, the LINEEYE protocol analyzers
check the number of received[Polling setup display]frames that are assigned theiraddress and whether errors
occur or not, replying withuser-set data. In the mastermode, they send polling
messages to 32 slave units, andcheck and display replies fromeach slave.
Only for LE-3500■ BUFFER mode

In the BUFFER mode, you can select between transmission and reception, and sendtransmitted or received data that has been captured in the buffer using the unit's
monitoring feature, as simulation data without requiring further manipulation. Thismode is effective in conducting[Buffer data reproduction setup display]
reproducibility tests using thesame data as that monitored
under actual communications
conditions.

■ MANUAL mode
The MANUAL mode allows you to send the data registered in transmission table
which corresponds to the "0" to "F" keys. The data can be sent with one press of a
key. While checking replies from a unit under development with the monitoring
feature, you can easily and simply test the communications process. You can also
send fixed data by registering it[Transmission table setup display]under a key combination of theSHIFT and "0" to "D" keys, aswell as turn RTS/CTS andDTR/DCD signal lines on/off
with the SHIFT and "E"/"F" keycombinations.

■ FLOW mode
Flow control can be simulated on the transmission and reception-lines using X-on/off
flow control or the control line handshake. In the transmission mode, up to 16 cyclesof data from transmission start until a generated interrupt request can be displayed. In
the reception mode, you can set[Flow control test setup display]
the number of received data
cycles until a transmissioninterrupt request is generated,
as well as the time until thetransmission resume request is
generated.

■ ECHO mode
In the ECHO mode, LINEEYE protocol analyzers internally return received data. Buffer
echo to send back data by a reception frame, character echo to send back data by acharacter and loop back echo
[ECHO mode setup display]that simply loops back data canbe selected. It is used to test
displayterminals
andcommunications terminals.

Only for LE-3500■ PROGRAM mode
By creating a purpose-specificcommandprogram,thecommunications protocol can be
flexibly simulated alongsidecondition monitoring. The programis created using the menu
selection, so it is easy to master.

[Program example][PROGRAM mode commands]
CommandOperation

SEND CHR□□□□□□□□

Sends max. 8 data sets.

SEND REG□

Sends data registered in transmission table under
specified REG No.

SEND BRK
WAIT

CHR

WAIT

FRM

WAIT

TM

Sends break signals (ASYNC only).

□□□□□□□□

Waits until receiving specified data (max. 8 data sets).
Waits until receiving 1 frame.

□□□□

GOTO L□□□

Waits for specified amount of time.
Jumps to specified label No.

CALL

L□□□

IF

CHR

□□□□□□□□IF

LN□＝□ L□□□

SET

REG

□

□□□□□□

Sets or increases/decreases value of specified REG
No.

SET

TM

□

□□□□□□

Controls specified timer and sets to specified value.

INT

TRG 0 L□□□

Jumps to subroutine of specified label No.

L□□□ Branches if specified data in reception buffer.
Branches if interface line is specified logic.

Interrupts specified label when trigger 0 condition issatisfied.

Easy-to-Use Handy Functions
Continue to Evolve
Firmware That Evolves

PC-compatible File Management Specification

The latest firmware with additional functions and improvements can be foundon our website. If you download it with your PC, you can then update to thelatest version via a serial/ USB cable.

Menu-based Simple Operation
Anyone can easily use LINEEYE protocol analyzers owing to the easy menu
selection system handed down from earlier models.

Test conditions and results such as measured data can be saved on optional CF
cards in the files management format compatible with your PC. Of course, files canbe interchangeably used (*) between models. Therefore, measurement data can besaved on-site with the LE-2500, and analyzed or manipulated in greater detail using
the LE-3500 back in the office.
* The LE-8200(A)/3500/2500/1500/7200/3200/2200/1200 are compatible in measurement data file. Partof files or data saved in higher hierarchy models or new models, however, may not be available to lower
hierarchy models or conventional models.

[Top menu display (LE-3500)]

Types, names, sizes and thedate/time of files saved in thememory card can be checked.

[Example file operation display]

When many files have beensaved, the file filter featureallows you to specify the typeof file to be displayed.

[File filter setup display][Top menu display (LE-2500)]

A backlit LCD makes it easy to view
measurement data at night and indark places.

Offline Analysis and Data Searches
Measurement data displays can be freely scrolled and paged. A powerfulsearch feature allows you to locate specific data and perform counting

Search key

Communications error (individual error type can be specified),communications data string of max. 8 characters (don’tcare and bit mask can also be specified), idle time beyond
a specified duration, specific time stamp (don’t care canalso be specified), external trigger matching data

Searchoperations

Find and display, counting

Auto RUN/STOP for Unmanned Measurement
By setting time and a date of measurement start and end, measurement can bedone automatically during the specified time period. For example,
measurement only for 3 hours from 18:00 to 21:00 every day is possible.
Besides, if the power ON auto run function is used, unmanned measurement
can be started automatically without pressing the RUN key after turning power ON.
[Example auto run display][Example search key setting]

Using “don’t care (*)”, you can search fortime stamp data from 10:30:00 to 10:39:59
as in this example.

Various Print Formats
Measurement data of a user-specified range can be
printed out continuously from any printer, in the text
format that corresponds to the display mode. Print
data output from the AUX (RS-232C) port can besaved as a text file in the PC using the utility
software or HyperTerminal.
And, with a dedicated printer, you can print hardcopy
of display images, continuous image of logic
analyzer waveforms, and results of statisticalanalysis.

[Example statistical analysis printout][Example logic analyzer waveform printout][Example print hard copy of display image]【Utility image】
Dedicated printerDPU-414-PA

Output fromthe AUX port

The printing image is saved as file.

The PC installed the utility software LEPRTIN_WIN(*)
to capture printout data.
* The software can be downloaded from LINEEYE's website.

LE-PC300G Enhances the Link
between Analyzers and your PC
PC Link Software

LE-PC300G
Enables simultaneous control of multiple analyzers from a PC
The LE-PC300G supports serial connections through the COM port, USB
connections, and LAN connections via LINEEYE LAN-Serial converter, thus enabling
remote measurement by multiple analyzers connected at the same time. It also
allows you to browse measurement data saved in memory cards and convert data.

[Linking multiple analyzers by serial and LAN connections]Target line

Target line
PC

【Serial connections and memory card】
Target line

PC with PC link software
installed

LE-3500

LE-2500*1

SI-60F

LAN（10/100BASE-TX）*1

Measurement data
Measurement data

SI-60

Target line

*1

Target line

SI-60

*2
*2
CF card*2*1

LE-2500*1: The PC Link software is not provided with a USB cable. Prepare a USB cable if you intend to use USB connection. The LE2-8V AUX
cable provided to the analyzer is available in the case of serial connection.
*2: An interface to read the CF card is required on the PC side.

LE-3500*1: SI-60/SI-60F is a LAN-serial converter supported by LE-PC300G. Target analyzer is identified by specifying IP
address of SI-60/SI-60F on the remote setting window of LE-PC300G.
*2: Optional AUX cable for DSUB 25-pin (LE2-8C). Set the DTE/DCE switch of SI-60 to DTE.

Allows the measurement data to be checked on your large PC screen.

▲ Normal display▲ HDLC translation display▲ Logic analyzer display

Records communication logs continuously on PC up to a maximum of 16GB

Converts the recorded data to text format or CSV format all at once

The remote monitor function allows to record the data measured by ananalyzer on the hard disk of PC.
The fixed buffer mode and ring buffer mode are available. The former stops
recording when the specified data size is reached, and the latter records data
endlessly within the limit of the specified size.

Multiple files of communications logs can be converted to text or CSV format
for use on word processor or spreadsheet. Conversion to text is based on theprint format of the analyzer. In consideration of analysis on general searchtool, it is possible to delete decorative guides or time data, and to specify
conversion of sender or receiver data only.

[Standard time for continuous recording on hard disk *1 ]Target line speed *2

When 1 GB is specified :(e.g.: 1 MB x 1,000 files)

When 16 GB is specified :(e.g.: 8 MB x 2,000 files)

9600 bps

Approx. 60 hrs

Approx. 960 hrs

19200 bps

Approx. 30 hrs

Approx. 480 hrs

38400 bps

Approx. 15 hrs

Approx. 240 hrs*1: In case of full-duplex communications line for transmission at 1 ms interval per 1 KB.
*2: Maximum communications speed that ensures recording of measurement data without failure will be about 1/5 of serialtransfer speed between analyzer and PC.

Changes the System Language Automatically
The system language alternates automatically between English and Japanese
according to that of OS. This facilitates introduction of the software todevelopment bases outside Japan.

■LE-PC300G Specifications
Applicable analyzers

LE-3500, LE-2500, LE-1500, LE-7200, LE-3200, LE-2200, LE-1200

Analyzer connection

Serial, USB (for full-speed transfer), and LAN (with SI-60/SI-60F unit sold separately)

No. of analyzers to be connected

Multiple analyzers can be connected and controlled simultaneously.(No. of connectable analyzers depends on the performance of PC.)

Key emulation function

Presents the analyzer’s display on the PC screen to enable control in a manner as if operating the analyzer.

Measurement condition setting

Measurement conditions (communications parameters, trigger and simulation data) can be input and edited on the dedicated window on PC screen.

Remote monitor function

Recording modes Fixed buffer mode (Records data up to the specified size) or ring buffer mode (Records data endlessly whileleaving the latest data of the specified size) can be selected.

PC Link Software for CAN/LIN
For OP-SB7GX LE-PC7GX

Starts/stops measurement with analyzer, displays the measurement data on PC screen, and records data continuously.

Recording capacity Max. 16 GB can be specified up to 2,000 files in the unit of 1/2/4/8 MB data file.
Selectable among raw data, protocol translation and logic analyzer waveform.

Raw data
Display modes

Displays communications data accompanied by idle time, time stamp and line status.
Character code (10 kinds) and character size (small/medium/large) can be changed.

Protocol translation Translates and displays SDLC, X.25 and LAPD protocols. (Target protocols planned to be increased.)
Logic analyzer waveform Enlarges and reduces waveform, measures time between cursors, and rearranges signals.
Display area

Display window size can be changed.

Character codes

ASCII, EBCDIC, JIS7, JIS8, Baudot, Transcode, IPARS, EBCD, EBCDIK, HEX (in hexadecimal including error codes)
Finds and displays the data that matches the search key.

Search function

Search key

Specified data string of max. 8 characters (don’t care and bit mask can also be specified), idle time beyond a specified duration, specific time stamp (don’tcare can also be specified), error (parity, framing, BCC, break/abort, short frame: individual error type can be specified) external trigger matching data

Text-CSV conversion function

Specified number of recorded files can be converted to text or CSV format all together.

Bitmap conversion function

Analyzer’s display shown by key emulation can be saved to bitmap files.

System requirements
CompositionPC

PC / AT compatible CPU: Pentium3 1GHz or faster
RAM: 256 MB or more (recommended)
HDD: 5 MB + free bytes on the measurement data area

OS

Windows ® 2000 / Vista ® / 7 / 8

CD (Software) x 1, instruction manual x 1, user registration card x 1

This software links your PC and LE-3500/2500 equippedwith CAN/LIN communications expansion kit OP-SB7GX.
It enables to analyze collected CAN/LIN data on your PC.

••
••
••
••

USB, Serial and LAN connection to the PC
Key emulation function for remote control
Recode CAN/LIN data into the PC at maximum 16GB
Display the specific ID frame at real time
Data and timestamp search, text/CSV conversion
Set the analyzer conditions from the software
Read the measured file on the CF card
OS: Windows® /Vista®/7/8

Expanded communications

TTL/I2C/SPI IrDA/ASK CAN/LIN CC-Link

Optional measurement boards along with a variety of measurement
cables expand the application range of the LE Series.

▲New communications standards different in ▲Cables and terminal adapters in a wide variety
hardware specification are supported with theare available to meet the shapes of thereplacement of the measurement board.
connectors of measurement targets.

TTL/I 2 C/SPI Communications Expansion Kit SB5GL
This interface expansion kit can measure RS-232C (V.24) and TTL/C-MOS signal levels used
between LSI chips on printed circuit boards (PCB). TTL/C-MOS port supports monitoring andsimulating the I2C/SPI (*1)communications besides UART/HDLC communications at 1.8V to 5V.
Furthermore, it supports BURST mode, which captures all data synchronizing clock signals.

SDA
SCL

* I2C (Inter-Integrated Circuit) is a
serialinterfacestandard
for
peripheral devices developed byPhilips Corporation.

[I2C protocol setting screen]

LE-3500, LE-2500, LE-1500 (*2)
RS-232C, TTL / CMOS (for I 2C and SPI (*1))
SD (SDA/SDO) , RD (SDI) , RS (SS) , CS , EX IN , SD CLK (SCL/SCK), RD CLK ,Trigger IN , Trigger OUT [Lead length: 170mm]
I 2C, SPI (*1), BURST
Monitor / Simulation / BERT (*3)
SPI : 50bps to 2.048Mbps (*4) , 115.2Kbps to 10Mbps on using OP-FW10G (*5)
I2C : max. 1Mbps (On simulation 50K, 100K, 200K, 384K, 417K, 1Mbps)
5.0V/ 3.3V/ 2.5V/ 1.8V signal level (selectable)
5.0V setting: High: Min 3.5V, Low: Max 1.5V
3.3V setting: High: Min 2.0V, Low: Max 0.8V
2.5V setting: High: Min 1.7V, Low: Max 0.7V
1.8V setting: High: Min 1.2V, Low: Max 0.6V
Dedicated expansion board, relay cable, probe pod, probe unit

Applicable analyzers
Interface
Probe signal[Example of I2C monitor display]

ProtocolTest function
Baud Rate
TTL/CMOS signal level
Input Level
Composition*1 : SPI is only available with LE-3500.
*2 : LE-1500 does not support I2C/SPI*3 : BURST monitoring only. BERT for I2C/SPI are not supported.
*4 : When the continuous transfer is less than 1Kb, Max 5Mbps at monitoring and Max 2.048Mbps at simulation.
*5 : Applied in monitor mode. Max 5Mbps at simulation mode.

Infrared Communications Expansion Kit OP-SB6G
This expansion kit is provided with a probe pod for monitoring IrDA and ASK infraredcommunications. The kit has an IrDA monitor function that makes it possible to change
communications speed automatically according to the IrLAP protocol and allows the seamlessmonitoring of infrared data, the mode of which changes from SIR (9600 bps) to FIR (4Mbps). The
kit has two optical emission levels (high and low levels), either one of which is selectable.

ᰓ
ᰑ

IR

IR

* IrDA is half-duplex wireless communications protocol that uses infrared rays.

ᰒᰑ IR receiving
window on SD side
ᰒ IR receiver on
RD side[Example of IrDA monitor display]

IR: Infrared rays

ᰓ The simulation
function enables theoutput of IR data fromthe analyzer. The
emission output canbe set to low or high.

Applicable analyzers
Interface
Measurement signal
ProtocolFunction

LE-3500, LE-2500
RS-232C, IR (IrDA/ASK)

Baud rate (bps)
Output emission level
Composition

Photodiode / LED : HSDL-3602 or equivalentSD,RD
IrDA1.1(SIR/MIR/FIR (*1)),ASK
Monitor/Simulation/BERT (*2)
2400, 9600, 19.2K, 38.4K, 57.6K, 115.2K, 0.576M, 1.152M (*3)
Automatically detects and follows IrLAP protocol. (*1)
High/low interchangeable
Dedicated expansion board, relay cable, probe pod*1: The Expansion Kit is in automatic tracking control of MIR (at 576 kbps or 1.152 Mbps) or FIR (at 4Mbps) while the expansion kit is in monitoring. Due to the performance of the analyzer, however,continuous data may not be correctly captured midway. *2: IR bit error rate testing (BERT) is not possible.
*3: Settings are not possible in the LE-2500.

Current Loop Adapter and Expansion Board OP-1C ＋ SB-25L
The OP-1C used in combination with the SB-25L (*1) supports current loop communications
presently used in the FA field. The kit incorporates a communications circuit with photocoupler
built in OP-1C insulation and constant-current power supply of insulated type, thus realizing not
only monitoring but also easy communications testing with passive or active current loop
devices.

Transmission: Active device
Reception: Passive device

Current direction
SO

SI

Note 1: The dedicated expansion board provided to the OP-SB5G/OP-SB5GL or OP-SB6G can be used in placeof the SB-25L. The purchase of the SB-25L is unnecessary if the dedicated expansion board is on hand.

[OP-1C Current Loop Adapter]

OP-1C

Current direction

+V

R

RI

RO

[SB-25L Expansion Board]* The current loop communications system is noise-resistant communications method
in which logic is decided with the ON/OFF of the current.
Applicable analyzers

LE-3500, LE-2500, LE-1500

Interface

RS-232C, Current loop communications (4-pole terminal block)

Measurement signal

SD, RD

Baud rate (bps)19.2 kbps max. (*1)

Function

Monitor/Simulation

Monitor current level

10〜60mA

Signal polarity

Normal/inversion (*2)

Simulation mode

Passive test and active test
Active current: 20/40 mA (selectable with DIP switch)

OP-1C dimensions and mass

60(W)×100(D)×20(H)mm, Approx. 180g

OP-1C accessories

Dedicated adapter and relay cable*1: The baud rate is restricted by the cable length and current value. *2: The signal polarity is set in theanalyzer. OP-1C is not provided with DIP switches for polarity switching equivalent to that of the conventionalmodel OP-1B.

Increases in the efficiency of developing and testing in-vehicle networks.
CAN/LIN Communications Expansion Kit OP-SB7GX
This expansion kit makes the measurement of up to 2 channels simultaneously byusing Controller Area Nerwork (CAN) communications used widely in FA systems andin-vehicle communications, and Local Interconnect Network (LIN) communications datain flexible connection. This expansion kit allows the simultaneous logic measurement
and analog measurement of four-line external signals.

●CAN/LIN Simultaneous Monitoring
The OP-SB7GX enables the simultaneous measurement of CAN
communications data and LIN communications data along with time stamp,
thus contributing to the development of bridge units connecting the CAN
and LIN. Furthermore, the ID filter can be used for highly efficient analysis.

LIN*CAN is the communications protocol for in-vehicle network
developed by Robert Bosch GmbH. LIN is communications protocol for
in-vehicle network proposed by automobile manufacturers in Europe.

CAN[Example of CAN/LIN monitor display]TM

Displays the time of frame reception completion in millisecond units (Example: 4216898 → Reception at 42 minutes 16.898 seconds)
The [ZOOM/CODE] key can be used to select the display of the difference in time (dT) from the moment the previous frame is received.

CH

Reception channels (1: CH1 and 2: CH2)ID

CAN: ID of received frame, LIN: Displays the following items in sequence. SynchBreak bit width, SynchFiled, "-" and ID(Example: 1355-35, SynchBreak width=13 bits, SynchField=55h, and ID=35h)

TYP
DL
DATA

Types of reception frames

DAT: CAN data frame

REM: CAN remote frame

ERR: CAN error frame

FRM: LIN frame ILL: Illegal frames beyond LIN standards
CAN: Contents of data length code (number of data bytes) LIN: Data length set for CONFIG for each ID
Contents of data fieldC

Contents of LIN checksum (hexadecimal)

S

Displays whether or not the frame was normal.

I

Displays the logic of external signal IN1. All the external signals will be displayed with the [DISPLAY MODE] key.

●CAN Simulation Function
A frame registered in the CAN data table is transmitted. A partof the data in the frame can be specified as sweep data thatcan be transmitted with the value of the data automaticallychanged from the initial value to the third target value, whichmakes it easy to check the response of the equipment
according to the change of communications data.

[Settings for CAN data table][Image of sweep data settings]
DATA

Sixteen frames specified with the number of repeating times
and transmission intervals can be registered in advance.

TIME●LIN Simulation Function
The OP-SB7GX in master mode can transmit the contents of
the LIN data table in the order set in the schedule tablerepeatedly or according to key manipulation. A parity error, any
number of break bits, and any SYNC data can be set to
conduct confirmation tests for error data with ease. While inslave mode, the contents of the data table set with an ID
conforming to the request of the master will be transmitted.
Furthermore, the WakeUp signal (80h) can be transmitted at
any time.

Applicable analyzer[Example of master mode settings]

A 15-bit response space (the space between the header
block and response block) and 5-bit inter-byte space (thespace between adjacent response data items) are set.

CAN: Conforms to ISO11898/ISO11519-2 standards (*1) (DSUB9-pin connectors x 2)

Transceiver

CAN: Equivalent to TJA1050/1054

No. of measurement channels

2 channels in total with CAN, LIN or CAN / LIN in combination

ProtocolCAN2.0B, device net, LIN (Rev 1.1, 1.2, 1.3, and 2.0, 2.1)

Monitor function
Trigger function
Simulation function
External signal input
Composition

Data table numbers 0, F, 1, and 2 are transmitted insequence with a parity error set for an 18-bit-length BREAK
filed for table F data.

LE-3500, LE-2500

Interface

Baud rate (bps)[Example of schedule table settings]

CAN: 1 Mbps max.

LIN: Conforms to ISO9141 standards (header 3-pin connector x 2)

LIN: Equivalent to TJA1021

LIN: 26 kbps max.

Arbitrary baud rate settings possible
ID filter possible and time stamp (1 ms min.) recordable
CAN: Standard/Expansion format supported and possible to make bit timing settings

LIN: Frame breaking possible according to the data length of each ID or specified idle time

Conditions: Specified communications data string (8 characters max.), specified remote frame (CAN), frame error (LIN), timer and counter coincidence, and external signal logic
Operation: Measurement stop, saving in a memory card, timer control, counter control, specified data transmission, buzzer, and validation of trigger conditions
Pre-registered data is selected by key operation and transmitted (sweep transmission available) and Master and slave simulation (LIN only).
Real-time display of 4-channel external signal state with LED possible.
Signal logic recordable in synchronization with data continuous measurement of voltage possible. (Measurement range: ±15 V)
Dedicated expansion board, DB9 monitor cable x 2, 3-wire probe cable x 2, and 8-wire probe cable*1: Settings in the Analyzer selectable with a relay.

High-speed HDLC/SPI Communications Firmware

OP-FW10G

This expansion firmware increases the baud rates of bit
synchronous communications (e.g., HDLC/SDLC/X.25,
and CC-Link communications)and SPI communications
up to 10 Mbps. The firmware processes main measurement
items completely with a field programmable gate array(FPGA), thus precisely capturing communications data
along with time stamps in 1-µs units. It is useful to measure
the high-speed HDLC communication at CC-LINK of
RS-485 multi-drop type, and the high-speed SPI/HDLC
communications at TTL signal level on the PCB boards.

Applicable analyzer

LE-3500

Interface

RS-422/RS-485 (RS-530) (*1), TTL(*2), SPI(*2)

ProtocolHDLC, SDLC, X.25. CC-Link (NRZ/NRZI format, AR clock), SPI

Baud rate (bps)Error check
Monitor function

Simulation function[Screen for HDLC settings]

LE-25TB

RS-485PLC

Half-duplex 115.2 kbps ࠥ 10 Mbps (*3)
Full duplex

115.2 kbps ࠥ 5 Mbps (*3)

FCS error (CRC-ITU-T), abort, short frameTime stamp 6 digits (0 to 524287) Resolution: 1 ms, 100 µs, 10 µs, or 1 µs (selectable)
ID filter (HDLC) Specify 2 characters (don't care and bit mask available)
Data table

16 kinds up to 16k in total

MANUALmode

Data can be transmitted by key operations.
Able to set repeat mode and idle time.

Trigger function

When 2 individual or sequential characters, errors or the external trigger input (lowlevel) are found, the analyzer automatically stops monitoring.

Data search

Search trigger data, error data or any data strings.

Composition

Firmware CD, instruction manual.

*1: LE-25TB and LE-530TB are useful to connect to the target device.
*2: OP-SB5GL is required.
*3: OP-SB5GL is required for high-speed simulation of TTL/SPI.

LE-3500 / LE-2500 Specifications
LE-3500

LE-2500

RS-232C （V. 24） (*1)◎◎

RS-422/485 （RS-530） (*1)

◎

Model
Interface

Expansion measurement
interface.

Expansion firmware

Standard Protocol

Optional Protocol○ [ LE-25Y15 ]○ [ LE-25Y15 ]RS-449○ [ LE-25Y37 ]○ [ LE-25Y37 ]V. 35○ [ LE-25M34 ]○ [ LE-25M34 ] (*2)

TTL/I 2C/SPI (*3)

○ [ OP-SB5GL ]○ [ OP-SB5GL ]Infrared communications IrDA/ASK○ [ OP-SB6G ]○ [ OP-SB6G ]Current loop

○ [ OP-1C + SB-25L ]○ [ OP-1C + SB-25L ]CAN/LIN○ [ OP-SB7GX ]○ [ OP-SB7GX ]High-speed HDLC/CC-Link○ [ OP-FW10G ]−ASYNC (Asynchronous), ASYNC-PPP◎◎

Character synchronous SYNC/BSC◎◎

Bit synchronous HDLC/SDLC/X.25◎◎

Modbus◎

Not supported

I 2C

◎

◎

SPI

◎

Not supported

BURST (*4)◎◎

IrDA(IrLAP)◎◎

CC-Link◎

Not supported

CAN

◎

◎

Devicenet

◎

◎

LIN

◎

◎

ST1 (DTE transmission clock), ST2 (DCE transmission clock), RT (DCE reception clock),
AR (The synchronous clock extracted from the edge of the transmission and reception data)

Synchronous clock
Memory capacity (*5)
Capture memory

6.4 MB

2.4 MB

Battery backup

Approximately 5 years with built-in lithium battery

Added function with memory used

Baud rate◎

X. 20/21

Two divided areas, data protection, and selection between fixed-size buffer and ring buffer.

Max. speed (full-duplex)

1.544Mbps

Max. speed (half-duplex)

2.048Mbps

1.000Mbps

50bps〜2.048Mbps

50bps〜1.000Mbps

Speed setting range
Speed setting step, accuracy

1.000Mbps

Freely set to four effective digits, separately for transmission and reception.(Margin of error: ±0.01% or less)

Data format

NRZ, NRZI, FM0, FM1, 4PPM, ASK

Data code

ASCII, EBCDIC, JIS7, JIS8, Baudot, Transcode, IPARS, EBCD, EBCDIK, HEX
Asynchronous

Character Framing

Data bit (5, 6, 7, 8) + parity bit (0, 1) + stop bit (1, 2)

Character synchronous

Data bit + parity bit (6 or 8 bits in total)

Bit-oriented synchronous

Data bit (8 bits)
NONE, ODD, EVEN, MARK, SPACE

Parity bit

MP (multiprocessor) bit is shown with a special mark.

Multi-processor bit

LSB first or MSB first (switchable)

Bit transmission order

Normal or Inverted (switchable)

Polarity inversion
Error check

Parity (ODD, EVEN, MARK, SPACE), Framing, Break, BCC (LRC, CRC-6, CRC-12, CRC-16, CRC-32, CRC-ITU-T).
BCC permeation mode.

For all protocols
For bit-oriented synchronous protocol
Specification

Abort, short frameCommunication log is recorded continuously and displayed in the LCD without affecting the communication lines.
OFF (no recording); Resolusion: 100ms, 10ms, 1ms; Max 999. 9 sec

Idle time display
Time stamp display

Date time stamp: Unit selectable among "Day/Hr/Min," "Hr/Min/Sec," "Min/Sec/10ms," or OFF (no recording)Line status display

Records and displays the wave form of 4 signals (chosen from RS(RTS), CS(CTS), ER(DTR), DR(DSR), CD(DCD),
CI(RI), EXIN(external trigger input) along with the transmission/ reception data.

Online monitor functions
Address filter

Records only frames of the specified address. (only when HDLC/SDLC/X.25)
Pause in capture, scroll, paging, jump to the specified screen.

Data display and operations
Bit shift display

Entire frame can be shifted to the right or left in 1 bit increments.
SDLC (modulo 8/128), ITU-T X.25 (modulo 8/128), LAPD, PPP, BSC, IrLAP, I2C

Protocol translation display
Target signals
Line status LED

Two color LEDs of SD, RD, RS(RTS), CS(CTS), ER(DTR), DR(DSR), CD(DCD), CI(RI), ST1(TXC1), ST2(TXC2),
RT(RXC).

RS-232C

Logic ON (red) , logic OFF (green) , no connection NC (light off)

Other interface

Logic ON (red) , logic OFF or no connection NC (light off)
2kinds; Max. count: 999999 (Resolution: 1ms ,10ms ,100ms)

Interval timer

2kinds; Max. count: 999999

General-purpose counter

For SD and RD (1 each): Max. count: 4294967295

Data counter
Simultaneous detection conditions

Up to 4 pairs of trigger condition and action can be specified. Sequential actions, which validate another condition after one condition is satisfied, is also possible.

Trigger condition

Communication error (Parity, MP, framing, BCC, break, abort, short frame can be specified individually.),
communication data string up to 8 characters (don't care and bit mask available), idle time more than the specified
duration, match time/counter value, logic status of interface signal line and extarnal trigger input

Trigger action

Stops measurement/test (offset can be set), validates trigger condition, controls timer (start/stop/restart), controls counter(count/clear), activates buzzer, saves monitor data on a memory card, sends the specified character string (during manual
simulation), and sends pulse output to external trigger terminal OT2.

External trigger output

Sends pulse to external trigger terminal OT1 when all conditions are satisfied. Sends pulse to external trigger terminal OT2 according to the trigger output specification.

Specification

Retrieves the data with specific condition from capture memory.

Search condition

Communication error (parity, MP, framing, BCC, break, abort, short frame), communication data string up to 8 characters (don't care andbit mask available), idle time more than the specified duration, specified timestamp, and trigger matching data.

Search action

Shows the match data at the top or enumeration display (selectable)

Trigger function

Data search function

Monitor conditions auto setting

Measurement conditions such as protocol, transmission speed, (max. 115.2Kbps), data code, synchronous character and BCC check can be set.

Auto run/stop function

Enables measurement to start and end at the specified time at the selected repeating cycle (monthly, daily, hourly).
Enables measurement to start automatically after power is turned ON.

Power ON auto run function

Auto save function

Specification

Automatically saves the monitored data in the capture memory and saves as communications log file in the CF card.

File size

BUF (capture memory size) , 1MB , 2MB , 4MB , 8MB , 16MB

Max. files1024LE-3500

Model

Measures and displays the value of voltage amplitude: SD, RD, ER (DTR), and CD (DCD) over RS-232C(current/min/max, range ±15V, resolution 0.1V).

Signal voltage measuring function

Takes statistics and displays graphs of transmission/reception data
count, number of frames, and satisfied trigger condition count.

Statistical analysis function

Logic analyzer function

LE-2500

Measures and displays the interval of change in the interface signal line. (current/min/max/average, resolution: 0.1ms)

Delay time function

Specification

Measures the logical change of the interface signal in the sampling clock period, and displays its wave.

Sampling clock

1KHz to 20MHz (14 steps)

Sampling memory

Min 2,000

Trigger condition

Trigger conditions in the ONLINE monitor functions match. Logical status of interface signal or external signal match.
Before, center, after

Trigger position

×8 , ×4 , ×2 , ×1 , ×1/2 , ×1/4 , ×1/8 , ×1/16 , ×1/32 , ×1/64

Zoom in/out
Other functions

Time measurement by cursor, signal line exchange, signal status searchSpecification

At DTE or DCE mode (selectable ), line quality measurement test such as error rates can be done by loop back test or interactive test.

Communication mode

Synchronous (SYNC), Asynchronous (ASYNC)

Measuring speed
BERT

Measurement mode(bit error rate test)

Test pattern

50bps~2.048Mbps, freely set to four effective digits

50bps~1.000Mbps, freely set to four effective digits

Continuous measurememt, specifies the number of receiving bit, specifies the time to measure, repeatedly measurement at the unit of 1 - 1440minutes
26-1, 29-1, 211-1, MARK, SPACE, ALT, DBL-ALT, 3in24, 1in16, 1in8, 1in4

Error bit insertion

Inserts 1-bit or 5-bit error in test pattern by key operation.

Measurement range

It is able to measure the parameter of the ITU-T advice G.821. It is able to output the external trigger by detecting the error bit.
Effective received bit (0〜9999999), bit errors (0 to 9999999 to 9.99E9), bit error rate(0 to 9.99E-9 to 1),
block errors (0 to 9999999 to 9.99E9), block error rate (0 to 9.99E-9 to 1), Savail(available measurement time: 0 to 9999999sec),loss count (synch loss: 0 to 9999), error duration (0 to 9999999sec), %EFS (normal operation rate: 0.000 to 100.000%)

Specification

Simulation function

Not available

Enables transmission/reception test of any given data in DTE or DCE mode (selectable).

Transmission data entry

Can be registered in 16 types of transmission data tables (Total of 16K data).

Error data entry

A part of transmission data can be registerd as error data such as parity error.

Line control mode

Auto Controls transmission timing with RS(RTS), CS(CTS), ER(DTR), CD(DCD) signal lines automatically in 1 ms
increments or manual (key operation) can be selected.

Transmission driver control

Auto control turning ON driver only during data transmission or manual mode linking with ER (DTR) or CD (DCD) key
operation can be selected during simuration of RS-485.

MANUAL mode

(Manual test)

Sends the data assinged to operation keys each time a key is pressed, while checking communications status on the display. Can be used together with the trigger function.

FLOW mode(Flow control test)

Simulates the X-on /X-off control data and flow control procedures of RTS/CTS control line. (Sender and receiver selectable).

ECHO mode(Echo test)

POLLING mode

(Multi-polling test)

BUFFER mode

PROGRAM mode

File management function(Buffer transmission test)(Program simulation)

Sends the received data frame by frame (buffer echo), by data (character echo) or by loop back.
Simulates multi-polling communications procedures. (Slave and master selectable)
Reproduces transmission of selected data (SD or RD)
captured in memory by monitor function.

Not available

Creates a simulation program (Max. type: 4, Max steps: 512)
using the dedicated commands (37 types) to test
the communication procedure.

Not available

Specification

Measurement data and condition can be saved in the CF card. And the format of the data/condition can be used in the PC.

File types

Measurement data (.DT), all measurement conditions (.SU), trigger save data (TG SAVEnn.DT), and auto save data (#nnnnnnn.DT)

File operations

Normal file display, file display by specified type/created date basis, save, load, delete, delete all, and format

Max. capacity (*7)
Printout function
LCD

16 GB

Monochrome 240 x 64 dots with backlight

B-connector in device side. Transfer data in full-speed. Can be used with PC [LE-PC300G], Can be used to upgrade the firmware.

External power supply (*8)
Built-in battery

Provided AC adapter Input: 100 to 240 VAC at 50/60Hz
Nickel hydrogen battery (Model: P-19S), Battery operating time(*9) : About 8 hours, Battery Charging time: About 2.5hours

Temperature range

In operation : 0 to 40 degrees, In storage : -10 to 50 degrees

Humidity range

85% (RH) max.

StandardCE(class A), EMC(EN61326-1 : 2006)

Dimensions

210 ( W ) x 154 ( D ) x 38 ( H ) mm

Mass
Accessories

Monochrome 240 x 64 dots with backlight

Mini DIN8 pin connector. Communication speed: 9600bps to 230.4Kbps (6 steps)
Print out data, Can be used with PC [LE-PC300G], Can be used to upgrade the firmware.

AUX(RS-232C) port
USB2.0 port

8 GB

Specified range of measurement data can be continuously printed in format corresponding to the display mode. Displayed images can be printed to make hard copies.

About 790g

About 760g

Monitor cable for DSUB 25-pin (LE-25M1), AUX cable for DSUB 9-pin (LE2-8V), external signal I/O cable(LE-4TG),
AC adapter(3A-183WP09), carrying bag(LEB-01), Utility CD, instruction manual and warranty◎: Standard support. ○: Supported with option product in [ ].
*1: An optional monitor cable (LE-259MI) and terminal block (LE-25TB/LE-530TB) are required in the case of monitoring over RS-232C with a D-sub 9-pin connector or RS-422/485 with a unique terminal arrangement.
*2: V.35 control signal lines are not supported. *3: SPI is onky supported by LE-3500. *4: Mode in which all data is imported in synch with clock edge. *5: Transmission/reception data, idle time, time stamp, and line
status consume 4 bytes of memory at each capture. *6: Correct auto settings are impossible if the amount of communications data is small or communications data includes a large number of errors. *7: Operation is
not guaranteed with memory cards not specified by LINEEYE. *8: The provided AC adapter (3A-183WP09 with a positive center plug) or the conventional AC adapter (FRA018-S09-U with a negative center plug) can beused. *9: When LCD back-light is OFF.

Order Information● LE-3500/LE-2500 ……………… (Comes with Japanese manual.)● LE-3500-E/LE-2500-E ………… (Comes with English manual.)Standard Set● Portable communication analyzer ........................ 1● DSUB 25-pin monitor cable (LE-25M1) ................ 1● DSUB 9-pin AUX cable (LE2-8V) ......................... 1● External signal I/O cable (LE-4TG) ...................... 1● AC adapter (3A-183WP09) .................................. 1● Carrying bag (LEB-01) ........................................ 1● Utility CD ............................................................ 1● Instruction manual ............................................... 1● Warranty ............................................................. 1

An easy-totransport carrying
bag is provided.

Options for LE-3500 LE-2500● Cables / Terminal blocks / Converter
Monitor cable for DSUB 25-pin

Monitor cable for DSUB 9-pin

Terminal block for DSUB 25-pin

LE-25M1

LE-259M1

LE-25TB

Branch cable for monitoring communicationlines over general DSUB 25-pin.

Branch cable for measuring RS-232C over
DSUB 9-pin of PC, etc.

Converts analyzer's RS-485/422 port (DSUB 25pin specification) to terminal block specification.

1.5m

0.1m

DB25(Male)

1.5m

DB25(Male)

DB25(Male) DB25(Female)

DB25
1
2
3

0.2m

Terminal block
1
2
3

DB9(Female) DB9(Male)
25*Same as the cable packed with analyzer.

25

X.21 Monitor cable

RS-449 Monitor cableLE-25Y15

LE-25Y37

Terminal block for RS-530

LE-530TB

Branch cable for measuring X.20/21 over
DSUB 15-pin. (Shield type)

Branch cable for measuring RS-449 over
DSUB 37-pin. (Shield type)

Converts TXD/RXD/GND signals on RS-530
port into terminals.

1.2m

1.2m
RS-530DB15(Male)
DB25(Male)

DB25(Male)
DB15(Female)

Terminal block

RXD− 3
RXD+ 16
GND 7
TXD− 2
TXD+ 14

DB37(Male)
DB37(Female)

1
2
3
4
5

V.35 Monitor cable

RS-530 cable

AUX cable for DSUB 9-pin

External signal cableLE-25M34

LE-25S530

LE2-8V

LE-4TG

Branch cable for measuring
V.35 over M34-pin.

A twisted pair cable
RS-530(shield type)

Cable for connection AUX (RS-232C)
port of an analyzer with PC (DSUB
9-pin DTE specification).

Probe cable for inputing/ outputing
external signal.

for*Same as the the cable packed withanalyzer.

• Length:2.5m

1.5m
1.5m

*Same as the cable packed with analyzer.

M34(Male)DB25(Male)

DB25(Male)

M34(Female)

DB25(Male)● Memory card

● Carrying bag

16G byte CF card

8G byte CF card

CF-16GX

CF-8GX

LEB-01

16G byte compact flash card, the operation of which
has been confirmed on LINEEYE's Analyzers.

8G byte compact flash card, the operation of which has
been confirmed on LINEEYE's Analyzers.

Bag with pockets for storing and carrying
accessories such as AC adapter, cables, etc.

Applicable model: LE-8200(A) and LE-3500

Applicable model: LE-8200(A), LE-3500, LE-2500, LE-1500*Same as the carring bag packed with analyzer.

● AC Adapter

Carrying bag● Battery pack

Wide input AC adapter

NiMH battery pack for replacement

3A-183WP09

P-19S

Input: AC100-240V, 50/60Hz
Output: DC9V, 2A
Plug: center

Rating: 4.8V, 1900mAH
Applicable model:LE-3500, LE-2500, LE-1500,
LE-7200, LE-3200, LE-2200,
LE-1200*Same as the AC adapter packed with analyzer.

Compact thermal Printer RoHS

*An auxiliary and replacement battery equivalent to
the Analyzer built-in battery.

Options
Directive
Compatible

AC adapter for DPU-414

PW-C0725-W1-U

Compact thermal printerCompact Thermal Printer Set

DPU-414-41B-E

DPU-414-PA

Built-in battery, dedicatedpaper (x1) included.

Includes printer (DPU-414-41B-E),
roll paper x1, AC adapter, and printercable (LE2-8P).

roll*AC adapter and cable are notprepared. Provide them separately.

Handy thermal printer for on-site
printout of measurements• Prints 40 digits per line in normal mode and 80 digits in reduced mode.
• High-speed printing at 52.5 characters per second.
• Incorporates eco-friendly NiMH battery.
• Supports Centronics parallel and RS-232C ports.
• Dimensions: 160(W)x 170(D)x 67(H)mm• Weight: Approx. 690g (including built-in NiMH battery)

Input: AC100Vࠥ240V
Output: DC7.0V, 2.5A(center)

Roll paper

TP-411L
Thermal roll paper for DPU-414.
10 rolls per carton.
Width: 112mm Length per roll: Approx. 28m

MULTI PROTOCOLANALYZER

LE-8200A/
LE-8200

240（W）×190（D）×48（H）mm,about1.1kg

Battery pack for DPU-414

BP-4005-E
Same as NiMH battery built-in
DPU-414-41B-E.
4.8V, 1100mAh

The top-level model of battery-powered portable communications analyzer with wide color display.
●Measurement at Low to Mega Speed up ●A long recording time of communications
to 4Mbps.
logs on 64GB CF card or USB flash drive (*1).
●100 Mbytes capture memory.
●Logic analyzer function and analog waveform analysis (*2).
●Supports to TTL, I2C, SPI, IrDA, CAN, LIN, ●A compact and lightweight model in B5 size
FlexRay, LAN and USB.
operating continuously for 4 hours.

AUX cable for DPU-414

LE2-8P
Cable for connection
AUX(RS-232C)port of analyzer
and serial port of DPU-414
Length:1.5m

Read the instruction manual provided with the product before use and use the productexplained in that manual. Using the product in ways not guaranteed in the manual,
SAFETY asconnecting
it to systems outside of the specified ranges and remodeling can all cause
and damage. LINEEYE CO. LTD. will assume no responsibility whatsoever for
WARNING trouble
trouble or damage arising because of unauthorized ways of use.

SisterProduct*1Only LE-8200A supports USB flash drive. *2High-speed analog waveform analysis requires an optional device.

• All brand names and product names mentioned in this catalog are trademarks or registered trademarks of theirrespective companies.
• Specifications and designs of products listed in this catalog are as of June 2014, and are subject to change
without notice for improvement.
• Colors of actual products may differ slightly from that listed due to printing condition.
• This catalog may not be reprinted or duplicated, in part or in whole.
©2014 by LINEEYE CO., LTD.

LINEEYE CO., LTD.
Head Office/Sales Office
Marufuku Bldg 4F, 39-1 Karahashi Nishihiragaki-cho, Minami-ku, Kyoto, 601-8468
PHONE: 81-75-693-0161 FAX:81-75-693-0163* LINEEYE CO. LTD. is a venture company founded by electronicequipment development members of the former Sekisui Chemical Co.,
Ltd. with investment from the Sekisui Venture Fund. The electronicequipment business of Sekisui Electronic Co. Ltd. was transferred to
LINEEYE CO. LTD. in October 2000.

QR-00337
ER-00094

L-14601E/LE G
Printed in Japan