AVAILABLE
with Signature
Calibration

SEE PG. IN-4

Ultra Precision
Wet/Wet Differential
Pressure Transducers
Model TJE

0.1% ACCURACY

1-888-282-9891

BI-DIRECTIONAMPLIFIED OUTPUT AVAILABLE
The Ultra Precision Wet/Wet Differential Pressure Model TJE is engineered for pressure
ranges of 50 to 750 psid with 0.1% full scale accuracy. This bi-directional transducer isdesigned to accept fluid in both ports. Durability is ensured for harsh industrial conditions
through standard features such as stainless steel construction and built-in overload protection.

Dimensions

www.honeywell.com/sensing

Sensotec Sensors

Model TJE (Order Code BD121)
Available Ranges*50; 75; 100; 150; 200; 300; 500; 750 psid
*Bold ranges are in stock.

Options (See Appendix)
Temperature compensated 1b, 1c, 1d, 1e, 1f; Internal amps 2b, 2n or 2N intrinsically safe amp see pageAP-6; 2j; 2k; Amp enhancements 3d; Pressure ports 5c; Electrical termination 6e, 6f, 6g, 6h; Int. shunt cal
8a; Signature calibration 53e
Premium Options: 1g, 1i; 2c; 5d; 6b, 6c, 6i, 6jAccessories: Mating connectors and connector/cable assemblies; Pressure port adapters

DP-18

1-888-282-9891

Model TJE
Order Code BD121
50 to 750 psid
±0.1% F.S.
2mV/V
Infinite

ENVIRONMENTAL

Temperature, Operating ...............
Temperature, Compensated ........
Temperature Effect- Zero (max) .............................
- Span (max) ............................

-65° F to 250° F
60° F to 160° F
0.25% F.S./100° F
0.25% Rdg./100° F

ELECTRICAL

Strain Gage Type .........................
Excitation (calibration) ..................
Excitation (acceptable) .................
Insulation Resistance ...................
Bridge resistance..........................
Shunt Calibration Data .................
Wiring Code (std) .........................
Electrical Termination (std) ..........
.....................................................
Mating Connector (not incl.) .........

Bonded foil
10VDC
Up to 15VDC or AC
10,000 megohm @ 50VDC
350 ohms
Included#2 (See Pg. AP-8)
PTIH-10-6P or equiv.
(Hermetic stainless)
PT06A-10-6S or equiv.

MECHANICALMedia............................................
Maximum Line Pressure ..............
Maximum Overload—
Safe (either side) ......................
Pressure Port(s) ...........................
Tapped Mounting Hole(s).............
Drain Port(s) .................................
Wetted Parts Material...................
Weight ..........................................
O-Ring Seals ................................
Dead Volume................................
Case Material ...............................

Gas, Liquid1500 psi
1500 psi
1/8-27 NPT Female
1/4-28 UNF X 5/16 DP
1/8-27 NPT Female
17-4 PH Stainless
4.6 lbs.
viton.25 cu. in
17-4 PH StainlessOutput Available ...........................
Additional Height ..........................

0-5VDC, 4-20mA
1.12"

INTERNALLY AMPLIFIEDUNITS (Optional)

ULTRA-PRECISION

Pressure Ranges..........................
Accuracy (min) .............................
Output (standard)*........................
Resolution ....................................

DIFFERENTIALPERFORMANCE*Notes * Unless otherwise specified on order, amplified units with 4-20mA output will provide 4mA at 0psid and 20mA at positive full scale and will not operate in the negative direction. An available
alternate set-up is 4mA at negative full scale and 20mA at positive full scale.
* 50 psid units have 1.5 mV/V output.

General Information
How to order (See Pg. AP-19)
Differential pressure selection flow chart (See Pg. DP-1)●お問い合わせは下記へ ̶DP-19