Configurable
Pressure Transducers
FP2000 Series
2-WEEK DELIVERY
mV/V, 0-5, 0-10 VDC, OR 4-20 mA

1-888-282-9891

GAGE, ABSOLUTE, VACUUM,BAROMETRIC
The FP2000 series is a manufacturing and delivery system which allows the customer to selectthe configuration which best fits the needs of the application. Choose from two accuracies, fouroutputs, six pressure ports, five electrical terminations and twenty-five pressure ranges. The FP
is available with gage, absolute, barometric or vacuum reference and, best of all, they deliver in
2 weeks or less.

Gage
FPG

Order CodeSensotec Sensors

PERFORMANCEENVIRONMENTAL

ELECTRICAL

www.honeywell.com/sensing

MECHANICAL

Pressure Ranges
Accuracy (BFSL)
Output (selectable)
Resolution

Absolute
FPA

VacuumFPV

See chart next page
0.1% or 0.25% F.S.
mV/V, 0-5 VDC, 0-10 VDC, or 4-20 mA (2 wire)
Infinite

Temperature, Operating
Temperature, CompensatedTemperature Error Band*
0.1% Accuracy
0.25% Accuracy-40° F to 240° F
40° F to 140° F±0.5% F.S.
±1% F.S.

Excitation (calibration)
Amplified (4-20mA, 0-5 VDC)
Amplified (0-10 VDC)
Unamplified (mV/V)

9 - 28 VDC
15 -28 VDC
10 VDC

Media
Overload-Safe
1000 psi and below
1500 psi and above
Wetted Parts Material

Gas, Liquid
4x FS or 3,000 psi, whichever is less
4x FS or 15,000 psi, whichever is less
Ha C276 & 316L ss*For ranges below 15psi, temperature effects may vary.
Non-amplified output @10 VDC excitation
FPG & FPA
0.10% accuracy
50 mV
0.25% accuracy
100 mVDimensions

FPV
33 mV
50 mV

FPB
40 mV
80 mV

CODE 6n
4-PIN, VENTEDMINI DIN

CODE 6a
6-PIN, VENTED
BENDIX STYLE

CODE 6q
4-CONDUCTORVENTED, INTEGRALCABLE, 5 FEET

GAGE, ABSOLUTE,BAROMETRIC & VACUUM.3".3".3"

CODE 6r
4-CONDUCTORVENTED, INTEGRALCABLE, CONDUITFITTING, 5 FEET

< 15 psi

PR-2

BarometricFPB< 5 psi

CODE 6m
4-PIN, VENTED
STD. DIN (43650).2".3"

1-888-282-9891
CODE 5a1/4-18 NPT Female

CODE 5b
1/4-18 NPT Male

CODE 5c
7/16-20 UNF Female

CODE 5d
7/16-20 UNF Male

CODE 5f
G1/4B Female

CODE 5g
G1/4 B Male

>1500PSI<1000PSI

Pressure
PortOptions

PRESSURE

www.sensotec.com/FP2000.htm

HOW TO ORDER

ExampleOrder Code:

TYPE
FPG

ACCURACY
1RANGEATOUTPUT
2u

PORT
5b

CONNECTOR
6aThis example is for a gage pressure unit, 0.10% accuracy, 5 psi range, mV/V output, 1/4-18 NPT Male
pressure port, and a Bendix electrical connection.
ELECTRICAL CONNECTIONS Bendix PTIH-106P (6a)
 MINI DIN, 40050 (6n)
 5´ Integral cable, polyurethane (6q)
 Std. DIN, 43650 (6m)
 Conduit fitting, (6r)

TRANSDUCER TYPE
 Gage (FPG)
 Absolute (FPA) Barometric (FPB) Vacuum (FPV)AVAILABLE OUTPUTS mV/V (2u)
 5 VDC (2d)
 10 VDC (2g)
 4-20 mA (2p)
 4-20 mA, CE (2y)
 4-20 mA, Intrinsically Safe (2n or 2N;
limited to ranges </= 5000 psi.)

ACCURACY 0.10% (1) 0.25% (2)GAGE/ABSOLUTE RANGES 10" H2O (WA)* 500 psi (CR)
 1 psi (AP)* 750 psi (CT) 5 psi (AT) 1000 psi (CV) 15 psi (BJ) 1500 psi (DJ) 30 psi (BM) 2500 psi (DM) 50 psi (BN) 5000 psi (DR) 100 psi (BR) 7500 psi (DT) 150 psi (CJ)
 10000 psi (DV) 250 psi (CN)*Gage only

ADDERS

BAROMETRIC
 16"-32" Hga (UQ) 26"-32" Hga (UR)
 0-30" Hga (UG)

VACUUM 10" H2Ov (WA)
 1 psiv (AP) 5 psiv (AT) 10 psiv (AV) 15 psiv (BJ)

PRESSURE PORTS
 1/4-18 NPT F (5a)
 1/4-18 NPT M (5b)
 7/16-20 UNF F (5c)
 7/16-20 UNF M (5d)
 G 1/4 B F (5f)
 G 1/4 B M (5g)

WIRING CODES, no shunt cal

Selecting an Adder will automatically
update the output code.

 Extended temperature range (1y) Buffered shunt cal (3d) CE rating (9e) Intrinsically safe, 2 wire (9d) See page AP-6
 CE and Intrinsically safe (9f) See page AP-6
 Potentiometers (14c)

OutputMatingConnectors

mV/V2u0-5 VDC 0-10 VDC
2d2g

4-20 mA
2p 2n or 2N

AA161 Mini DIN

#37

#38

#38

#54

AA111 Bendix

#2#50

#50

#49

#236q Integral Cable

#1

#52

#52

#51

#22

#53

6r Conduit

#1

#52

#52

#51

#22

AA157 Std DIN

#37

#38

#38

#54

#53

WIRING CODES, with shunt cal option (3d)ACCESSORIES

OutputMatingConnectors

Mating connectors with 15' of cable for Bendix connector (6a)
 mV/V 4-20 mA 0-5 / 0-10 VDCMating Connectors Only
 Mini DIN (40050)
 Bendix Standard DIN (43650)

2 WEEK DELIVERY

It’s easy to order exactly what you need. Simply make one selection in each of the required categories, chooseadders and accessories only if you want them. The result is a custom transducer with standard delivery!

without shuntwith shunt (3d)

AA113
AA116
AA117

AA513
AA516
AA517

AA161
AA111
AA157

mV/V2u0-5 VDC 0-10 VDC
2e2f

4-20 mA
2y 2n or 2N

AA161 Mini DIN

NA

#56

#56

#55

AA111 Bendix

#57

#60

#60

#58

#64
#59

6q Integral Cable

NA#63

#63

#61

#62

6r Conduit Fitting

NA

#63

#63

#61

#62

AA157 Std DIN

NA

#56

#56

#55

#64PR-3