SDG1000X
SeriesDataSheet-2022.06

Function/Arbitrary
Waveform Generator

SDG1000X Series Function/Arbitrary Waveform Generator

Key Features
Dual-channel, with bandwidth up to 60 MHz, and amplitude up
to 20 Vpp

SDG1062X
SDG1032X

150 MSa/s sampling rate, 14-bit vertical resolution, and 16 kpts
waveform length
Innovative EasyPulse technology, capable of generating lowerjitter Pulse waveforms, brings a wide range and extremely highprecision in pulse width and rise/fall times adjustment
Innovative TrueArb technology, based on a point-by-pointarchitecture, supports any 2 pts ~ 16 kpts Arb waveform with asampling rate in range of 1 μSa/s ~ 30 MSa/s
Special circuit for Square wave function, can generate SquareOverview
SIGLENT’s SDG1000X is a series of dual-channel function/arbitrary
waveform generators with specifications that include up to 60 MHz
maximum bandwidth, 150 MSa/s sampling rate and 14-bit verticalwaves up to 60 MHz with jitter less than 300 ps+0.05 ppm ofperiod
Plenty of analog and digital modulation types: AM, DSB-AM, FM,PM, FSK, ASK, PSK and PWM

resolution. The proprietary EasyPulse & TrueArb technique helps toSweep and Burst functions

solve the weaknesses inherent in traditional DDS generators whenHarmonics Generator function

generating pulse and arbitrary waveforms, and the special square
generator is capable of generating square waveforms up to 60 MHz
in frequency with low jitter. With these advantages, the SDG1000X
can provide users with a variety of high fidelity / low jitter signals
while meeting the growing requirements of a wide range ofcomplex and varied applications.

Waveform Combining function
High precision Frequency Counter
Standard interfaces: USB Host, USB Device (USBTMC), LAN (VXI11)
4.3” TFT-LCD display

WWW.SIGLENT.COM

SDG1000X Series Function/Arbitrary Waveform Generator

Models and Key Specifications
Product Model

SDG1062X

SDG1032X

Bandwidth60 MHz

30 MHzSampling rate

150 MSa/s

Vertical resolution

14-bitWaveform Length

16 kptsNum. of channels

2Max. amplitude±10 V

Display

4.3” display, 480 x 272 x RGB

Interface

Standard: USB Host, USB Device, LAN

Characteristics
Identical dual output-channels with high performance
Capable of outputting large signals at high frequencies. dual-channels, 20 Vpp amplitude can be guaranteed at up to 10 MHz.

Low Distortion Output
With 0 dBm output, the THD (Total HarmonicDistortion) is less than 0.15%. Harmonics andspurs are less than -40 dBc throughout the entire
bandwidth.

WWW.SIGLENT.COM

SDG1000X Series Function/Arbitrary Waveform Generator

Innovative EasyPulse Technology

When a Pulse waveform is generated by a common DDS generator, there will be a one-clock-jitter if the sampling rate is not an integer-related multiple of the output
frequency. SDG1000X EasyPulse technology successfully overcomes this weakness in DDS designs and helps to produce low jitter Pulse waveforms.

The rise/fall times can be set independently to the minimum of 16.8 ns at any frequency and to the maximum of 22.4 s. The adjustment step is as small as 100 ps.
The Pulse width can be fine-tuned to the minimum of 32.6 ns with the adjustment step as small as 100ps.

High performance Square Waves

Benefitting from a special square-wave generating circuitry, the Square from the SDG1000X breaks the 60 MHz bandwidth barrier, reaching rise/fall times of less than
4.2 ns, and frequencies up to 60 MHz.

The Square wave exhibits the same excellent jitter
performance as the Pulse waveform.

WWW.SIGLENT.COM

SDG1000X Series Function/Arbitrary Waveform Generator

Innovative TrueArb Technology
For arbitrary waveforms, TrueArb not only has all the advantages of traditional DDS, but also eliminates the probability that DDS may cause serious jitter anddistortion.

TrueArb generates arbitrary waveforms point by point, never skips any

As with EasyPulse, TrueArb effectively overcomes the defect that DDS

point so that it can reconstruct all the details of the waveform as defined.

may cause the one-clock-jitter in arbitrary waveforms.

Modulation

Burst

Multiple modulation types: AM, DSB-AM, FM, PM, FSK, ASK, PSK andTwo Burst modes, “N cycle” and “Gated”. The Burst source can be

PWM. The modulation source can be configured as “Internal” or “External”.

configured as “Internal”, “External” or “Manual”.

Sweep

Frequency Counter

Two Sweep modes, “Linear” and “Log”. Two Sweep directions, “Up” andHigh precision Frequency Counter with an input frequency range of“Down” and three Sweep sources, “Internal”, “External” and “Manual”.

0.1 Hz~200 MHz.

WWW.SIGLENT.COM

SDG1000X Series Function/Arbitrary Waveform Generator

Harmonics Function

Waveform Combining

Up to 16 harmonics may be generated. Amplitude and phase of each

Capable of combining the waveforms of 2 channels from internal,

harmonic can be set independently

providing more flexible tools to generate complex waveforms.

Arbitrary Waveform Software EasyWaveX

EasyWaveX is an arbitrary waveform software platform that supports waveform creation and editing. It features manual drawing, as-well-as line, equation, andcoordinate editing modes. It is also a convenient way for users to edit their own arbitrary waveforms.

WWW.SIGLENT.COM

SDG1000X Series Function/Arbitrary Waveform Generator

Specifications
All specifications apply to both channels. Unless otherwise stated, all specifications are not guaranteed unless the following conditions are met:
• The generator is within calibration period of validity• The generator has been working continuously for at least 30 minutes at a specified temperature (18℃ ~ 28℃ ).

Frequency Characteristics
ParameterMin.

Typ.

Resolution
Initial accuracy-25Max.

Unit

1μHzCondition+25ppm

1st year, 0~40℃

Max.

UnitCondition

60 M

Hz

SDG1062X

Sine Characteristics
Parameter

Min.

Frequency

1μTyp.

30 M
Harmonic distortion
Total Harmonic Distortion
Non-harmonic spurious

SDG1032X-60dBc0 dBm, 0~10 MHz ( included )

-45

dBc

0 dBm, 10~30 MHz ( included )

-40

dBc

0 dBm, 30~60 MHz

0.15%

0 dBm, 10 Hz ~ 20 kHz

-65

dBc

0 dBm, 0~10 MHz ( included )

-55

dBc

0 dBm, 10~30 MHz ( included )

-40

dBc

0 dBm, 30~60 MHz

Max.

Unit

Condition

60 M

Hz

SDG1062X

Square CharacteristicsParameter

Min.

Frequency

1μTyp.

30 M
Rise/fall times
OvershootDuty cycle

0.001

SDG1032X

4.2ns

10% ~ 90%, 1 Vpp, 50 Ω load

3.8

ns

10% ~ 90%, 2.5 Vpp, 50 Ω load

3%

100 kHz, 1 Vpp, 50 Ω load

99.999%Limited by frequency setting

300 ps +0.05 ppm ofperiod

Jitter (rms), Cycle to cycle

1 Vpp, 50 Ω load

Pulse Characteristics
Parameter

Min.

Frequency

1μPulse width

32.6

Pulse width accuracy
Rise/fall times

Max.

Unit

12.5 M

HzConditionns±(0.01%+1 ns)

16.8 nOvershootDuty cycle

0.001

Duty cycle resolution

0.001

Jitter (rms) cycle to cycle

Typ.

22.4

s10% ~ 90%, 1 Vpp, 50 Ω load , Subject to pulse
width limits

3%

100 kHz, 1 Vpp

99.999%Limited by frequency setting%
300 ps +0.05 ppm ofperiodps

1 Vpp, 50 Ω load

WWW.SIGLENT.COM

SDG1000X Series Function/Arbitrary Waveform Generator

Noise Characteristics
ParameterMin.

-3 dB bandwidth

60

Typ.

Max.

Unit

Condition

MHzRamp Characteristics
Parameter

Min.

Typ.

Max.

Unit

Condition

Frequency

1μ500 kHz

Symmetry

0100%1%Percentage of peak-peak output, 1 kHz, 1 Vpp, 50%

Max.

UnitCondition

6M

Hz

DDS modepts

DDS modeLinearity

Arbitrary Wave characteristics
Parameter

Min.

Frequency

1μWaveform length
Sampling rate
Vertical solution

16 k216 k

150 M
1μ30 M

14pts

TrueArb mode

Sa/s

DDS mode

Sa/s

TrueArb mode

bit
6.7

JitterTypes of built-in Arb

Typ.

ns

DDS mode, pk-pk

300

ps

TrueArb mode, cycle-cycle rms, 2 pts, 20.1 MSa/s

196

DC Characteristics
Parameter
Range
Accuracy

Min.

Max.

UnitCondition

-10

Typ.

10V

HiZ load-55V±(1%+3 mV)

50 Ω load
HiZ loadHarmonic Output Characteristics
Parameter

Min.

Typ.

OrderTypeMax.

Unit

Condition

16Even, Odd, All

Output Characterisics
Parameter

Min.

Max.

Unit

Condition

Range (Specified)(Note 1)

4m

20Vpp

≤ 10 MHz, HiZ load

4m

10Vpp

>10 MHz, HiZ loadRange (Setting)(Note 1)

2m20Vpp≤ 10 MHz, HiZ load

2m

10Vpp

Accuracy±(1%+1 mVpp)

Amplitude flatness-0.3

Output impedance

49.5

Output current-200

Crosstalk(CH1 - CH2 / CH2 - CH1)

Typ.

50+0.3dB50 Ω load , 2.5 Vpp, compare to 10 kHz sine,50.5

Ω10 kHz sine

200mA-60

dBc

Note 1: The specification will be divided by 2 when applied to a 50 Ω load.
WWW.SIGLENT.COM>10 MHz, HiZ load
10 kHz sine, 0 V offset

CH1= CH2= 0 dBm, Sine, 50 Ω load

SDG1000X Series Function/Arbitrary Waveform Generator

Modulation Characteristics
AMParameter

Min.

CarrierSine, Square, Ramp, Arb

Typ.

Modulation Source

Internal/External

Modulating waveSine, Square, Ramp, Noise, Arb

Max.

UnitCondition

Modulation depth

0120%Modulation frequency

1m20 kHz

While modulation source is “Internal”

Max.

UnitCondition

FMParameter

Min.

CarrierSine, Square, Ramp, Arb

Typ.

Modulation Source

Internal/External

Modulating wave

Sine, Square, Ramp, Noise, Arb

Frequency deviation

00.5*BW

Modulation frequency

1m20 kHz

While modulation source is “Internal”

Max.

Unit

Condition

BW is the max. output frequency limited byfrequency setting

Modulation Characteristics
PMParameter

Min.

CarrierSine, Square, Ramp, Arb

Typ.

Modulation Source

Internal/External

Modulating wave

Sine, Square, Ramp, Noise, Arb

Phase deviation

0360°

Modulation frequency

1m20 kHz

While modulation source is “Internal”

Max.

Unit

Condition

50 k

Hz

Limited by frequency setting while modulation
source is “Internal”

Max.

Unit

Condition

50 k

Hz

While modulation source is “Internal”

Max.

Unit

Condition

1M

Hz

While modulation source is “Internal”

ASK
Parameter

Min.

CarrierSine, Square, Ramp, Arb

Typ.

Modulation Source

Internal/External

Modulating wave

Square with 50% duty cycle

Keying frequency

1m

FSK
Parameter

Min.

Typ.

CarrierSine, Square, Ramp, Arb

Modulation Source

Internal/External

Modulating wave

Square with 50% duty cycle

Modulation frequency

1m

PWM
Parameter

Min.

Carrier

Pulse

Typ.

Modulation Source

Internal/External

Modulating wave

Sine, Square, Ramp, Noise, Arb

Modulation frequency

1mPulse width deviation resolution

6.67

ns

Burst Characteristics
Parameter

Min.

CarrierSine, Square, Ramp, Pulse, Noise, Arb

Type

Count(1-1000000cycles), Infinite, Gated

Carrier frequency

2mStart/Stop phase

0Internal period

1μTrigger source

Internal, External, Manual

Gated source

Internal/External

Trigger delay

Typ.

Max.

UnitCondition

BW

HzBW is the max. output frequency

360°

1000s

100s
WWW.SIGLENT.COM

SDG1000X Series Function/Arbitrary Waveform Generator

Sweep Characteristics
Parameter

Min.

CarrierSine, Square, Ramp, Arb

Typ.

Max.

UnitCondition

TypeLinear, Log

Direction

Up, Down

Carrier frequency
Sweep time

1μBW

HzBW is the max. output frequency

1m500sTrigger source

Internal, External, Manual

Max.

Unit

Frequency Counter Characteristics
Parameter

Min.

Function

Frequency, Period, Positive/Negative pulse width, Duty cycle

Coupling mode

AC, DC, HF REJ

Frequency range

Input amplitude

Typ.

Condition

100m200 M

HzDC coupling

10200 M

HzAC coupling

100 mVrms±2.5 V

DC coupling, < 100 MHz

200 mVrms±2.5 V

DC coupling, 100 MHz ~ 200 MHz

100 mVrms

5 Vpp

AC coupling, < 100 MHz

200 mVrms

5 Vpp

Input impedance

1M

AC coupling, 100 MHz ~ 200 MHz
Ω

Reference Clock Input/Output
Reference Clock Input
Parameter

Min.

Frequency

Typ.

Max.

10 M

Unit

Condition

HzAmplitude

1.4

Vpp

Input impedance

5kΩ

AC coupling

Reference Clock Output
Parameter

Min.

Frequency
Amplitude

2Output impedance

Typ.

Unit

Condition

10 M

Max.

Hz

Synchronized to internal reference clock

3.3

Vpp

HiZ load50Ω

Auxiliary In/Out Characteristics
Trigger Input
Parameter

Min.

Max.

Unit

VIH

2Typ.

5.5

V

VIL

-0.5

0.8

Input impedance

100

Pulse width

100

Condition

VkΩ
ns

Response time

100

ns

Sweep

600

ns

Burst

Trigger Output
Parameter

Min.

VOH

3.8

Typ.

Max.
0.44

VOL
Output impedance100

Frequency

Unit

ConditionV

IOH = –8 mA

V

IOL = 8 mA

Ω1

MHz

Sync Output
Parameter

Min.

VOH

3.8

Typ.

Max.
0.44

VOL

Unit

ConditionV

IOH = –8 mA

V

IOL = 8 mA

Output impedance100

ΩPulse width

500

ns

Frequency
Jitter (pk-pk)
WWW.SIGLENT.COM

106.7

MHznsSDG1000X Series Function/Arbitrary Waveform Generator

Auxiliary In/Out Characteristics
Modulation Input
Parameter

Min.

Frequency

0Input impedance

10Amplitude@ 100% Modulation
11depth

Typ.

Max.

Unit

50kHzCondition

kΩ
1213

Vpp

Typ.

Max.

Unit

Condition

2150

W

Dual channels, Sine, 1kHz, 10Vpp, 50Ω load

Typ.

Max.

Unit

Condition

General Characteristics
PowerParameter

Min.

Voltage

100 - 240 Vrms (± 10%), 50 / 60 Hz
100 - 120 Vrms (± 10%), 400 Hz

Power consumption

DisplayParameter

Min.

Color depth

24Contrast ratio

350:1

Luminance

300

bit
cd/m2

Environment
Parameter

Min.

Operating temperature

0Typ.

Max.
40Unit℃℃

Storage temperature-2060

Operating humidity

590%

550%

595%

Operating altitude

3048

m

Non-operating altitude

15000mMax.

Unit

Non-operating humidity

Condition≤ 30 ℃
40 ℃≤ 30 ℃

Calibration
Parameter

Min.

Calibration interval

Typ.
1

Conditionyear

MechanicalParameter

Min.

Dimensions

W×H×D = 260.3 mm×107.2 mm×295.7 mm

Typ.

Max.

Unit

Net weight

3.43kgGross weight

4.35kgCondition

Compliance
LVD

IEC 61010-1:2010

EMCEN61326-1:2013

Ordering Information
Product Description
60 MHz, 2 CH, 150 MSa/s, 14 bit

SDG1062X

30 MHz, 2 CH, 150 MSa/s, 14 bit

SDG1032X

Standard Configurations
Quick Start -1
Power Cord-1
Calibration Certificate -1
USB Cable -1
Optional Configurations
BNC Coaxial Cable

SDG-BNC

20 dB Attenuator

ATT-20dB

10W Power Amplifier

SPA1010WWW.SIGLENT.COM

SDG1000X
SeriesFunction/Arbitrary
Waveform Generator

About SIGLENTSIGLENT is an international high-tech company, concentrating on R&D, sales,
production and services of electronic test & measurement instruments.

SIGLENT ﬁrst began developing digital oscilloscopes independently in 2002.
After more than a decade of continuous development, SIGLENT has extended
its product line to include digital oscilloscopes, isolated handheld
oscilloscopes, function/arbitrary waveform generators, RF/MW signalgenerators, spectrum analyzers, vector network analyzers, digital
multimeters, DC power supplies, electronic loads and other general purpose
test instrumentation. Since its ﬁrst oscilloscope was launched in 2005,
SIGLENT has become the fastest growing manufacturer of digital
oscilloscopes. We ﬁrmly believe that today SIGLENT is the best value inelectronic test & measurement.

Headquarters:
SIGLENT Technologies Co., Ltd
Add: Bldg No.4 & No.5, Antongda Industrial
Zone, 3rd Liuxian Road, Bao'an District,
Shenzhen, 518101, China
Tel: + 86 755 3688 7876
Fax: + 86 755 3359 1582
Email: sales@siglent.com
Website: int.siglent.com
North America:
SIGLENT Technologies America, Inc6557 Cochran Rd Solon, Ohio 44139
Tel: 440-398-5800
Toll Free: 877-515-5551
Fax: 440-399-1211
Email: info@siglentna.com
Website: www.siglentna.com

Europe:SIGLENT Technologies Germany GmbH
Add: Staetzlinger Str. 70
86165 Augsburg, Germany
Tel: +49(0)-821-666 0 111 0
Fax: +49(0)-821-666 0 111 22
Email: info-eu@siglent.com
Website: www.siglenteu.com

Follow us on
Facebook: SiglentTech