SDS5000X Series
Digital Storage
Oscilloscope
Data Sheet
EN02E

Key Features◼

1 GHz, 500 MHz, 350 MHz models with real-time sample
rate up to 5 GSa/s◼

SPO technology

SDS5104X
SDS5054X
SDS5034X

The SDS5000X series employs Siglent’s SPO technology witha maximum waveform capture rate of up to 110,000 wfm/ s(normal mode, up to 500,000 wfm/s in Sequence mode), 256level intensity grading display function plus a color temperature
display mode. It also employs an innovative digital trigger
system with high sensitivity and low jitter. The trigger system
supports multiple powerful triggering modes including serial
bus triggering. History waveform recording, Sequenceacquisition, Search and Navigate functions allow for extended
waveform records to be captured, stored, and analyzed. An
impressive array of measurement and math capabilities,
options for a 25 MHz arbitrary waveform generator, as well as
serial decoding are also features of the SDS5000X.
The large 10.1’’ display capacitive touch screen supports multitouch gestures, with the addition of user-friendly one-button
design for most commonly used functions, can greatly improve
the operation efficiency of the SDS5000X. It also supports
mouse and external keyboard control.

WWW.SIGLENT.COM

Waveform capture rates up to 110,000 wfm/s(normal mode), and 500,000 wfm/s (sequence
mode)



Supports 256-level intensity grading and color
temperature display modes

Record length up to 250 Mpts/ch, 500 Mpts in total
for all 4 channels

Digital trigger system◼

Intelligent trigger: Edge, Slope, Pulse, Window, Runt,
Interval, Dropout, Pattern, Qualified, Nth edge,
Setup/hold, Delay and Video (HDTV supported). Trigger
zone helps to simplify advanced triggering◼

Serial bus triggering and decoder, supports protocols I2C,
SPI, UART, CAN, LIN, CAN FD, FlexRay, I2S, MIL-STD1553B, SENT, Manchester and ARINC429◼

Low background noise, supports 0.5 mV/div to 10 V/div
vertical scales◼

Segmented acquisition (Sequence) mode, dividing the
maximum record length into multiple segments (up to
100,000), according to trigger conditions set by the user,
with a very small dead time between segments to capture
the qualifying event◼

History waveform record (History) function, the maximum
recorded waveform length is 100,000 frames◼

Automatic measurement function on 50+ parameters,
supports statistics with histogram, trend, Gating
measurement, and measurements on Math, History andRef◼

Math function (2 Mpts FFT, addition, subtraction,
multiplication, division, integration, differential, squareroot, etc.), supports formula editor◼

Abundant data analysis functions such as Search,
Navigate, Digital Voltmeter, Counter, Waveform
Histogram, Bode plot and Power Analysis◼

High Speed hardware-based Average, ERES (Enhanced
Resolution); High Speed hardware-based Mask Test
function, with Mask Editor tool for creating user-defined
masks

◼

16 digital channels with sample rate up to 1.25 GSa/s,
record length up to 62.5 Mpts◼

25 MHz function / arbitrary waveform generator, built-in
multiple predefined waveforms◼

Large 10.1” TFT-LCD display with 1024 * 600 resolution;
Capacitive touch screen supports multi-touch gestures◼

Multiple interfaces: USB Host, USB Device (USBTMC),
LAN (LXI, VXI-11, telnet, socket, web), Pass/Fail, Trigger
Out, 10 MHz In, 10 MHz Out, VGA output◼

Built-in web server supports remote control by the LAN
port using a web browser; Supports SCPI remote control
commands; Supports external mouse and keyboard

Product Overview
SIGLENT’s SDS5000X series Digital Storage Oscilloscopes
are available in bandwidths of 1 GHz, 500 MHz and 350 MHz,
have a maximum sample rate of 5 GSa/s, maximum record
length of 250 Mpts/ch, and display up to 4 analog channels +16 digital channels mixed signal analysis ability.

1

SDS5000X Series Digital Storage Oscilloscope

Models and Key Specifications

Model

SDS5034X

Analog channels
Bandwidth
Sample rate (Max.)
Memory depth(Max.)
Waveform capture
rate (Max.)

4 + EXT
350 MHz
500 MHz
5 GSa/s (interleaving mode*), 2.5 GSa/s (non-interleaving mode**)Trigger type
Serial trigger anddecodeMeasurement
MathData analysis
Digital channel
Waveform
generator
I/O
Probe (standard)
Display*

SDS5054X

SDS5104X
1 GHz

250 Mpts/ch (interleaving mode), 125 Mpts/ch (non-interleaving mode)
110,000 wfm/s (Normal mode);
500,000 wfm/s (Sequence mode)
Edge, Slope, Pulse width, Window, Runt, Interval, Dropout, Pattern, Video, Qualified, Nth edge, Setup/hold,DelayStandard: I2C, SPI, UART, CAN, LIN
Optional: CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT, Manchester (decode only), ARINC429 (decode
only)
50+ parameters，statistics, histogram, trend supported
2 traces
2 Mpts FFT, +, -, x, ÷, ∫dt, d/dt, √, Identity, Negation, Absolute, Sign, ex, 10x, ln, lg, Interpolation, etc.;
supports formula editor
Search, Navigate, History, Mask Test, Digital Voltmeter, Counter, Waveform Histogram, Bode plot andPower Analysis
16-channel; maximum sample rate up to 1.25 GSa/s; record length up to 62.5 Mpts
Single channel external USB waveform generator, frequency up to 25 MHz, 125 MSa/s sample rate, 16
kpts waveform memory
USB 2.0 Host, USB 2.0 Device, LAN 10M/100M, Pass/Fail, Trigger Out, 10 MHz In, 10 MHz Out, VGA
Output
SP3050A, 500 MHz, 1 probe supplied for each channel
10.1” TFT-LCD with capacitive touch screen (1024*600)

Interleaving mode: only one of C1/C2 and/or only one of C3/C4 activated** Non-interleaving mode: both C1/C2 and/or both C3/C4 activated2

INT.SIGLENT.COM

SDS5000X Series Digital Storage Oscilloscope

Functions & Characteristics
10.1” TFT-LCD Display with Capacitive Touch Screen

10.1” display with 1024*600 resolution
Capacitive touch screen, supporting multi-touch gestures, can move or scale the waveform traces quickly by finger-touch
movements, which greatly improves the operation efficiency

Up to 110,000 wfm/s Waveform Update Rate

Record Length of up to 250 Mpts/ch

With a waveform update rate of up to 110,000 wfm/s, theoscilloscope can easily capture unusual or low-probability
events. In Sequence mode the waveform capture rate can reach
500,000 wfm/s

Using hardware-based Zoom technique and record length of up
to 250 Mpts, users can select a slower timebase withoutcompromising the sample rate, and then quickly zoom in tofocus on the area of interest

INT.SIGLENT.COM

3

SDS5000X Series Digital Storage Oscilloscope

Measurements of a Variety of Parameters

Parameter Statistics Function

Parameter measurements includes 4 categories: horizontal,
vertical, miscellaneous and CH delay providing a total of 50+
different types of measurements. Measurements can be
performed within a specified gate period. Measurements on
Math, Reference and History frames are supported

Statistics shows the current value, maximum value, minimum
value, standard deviation and mean value of up to 12
parameters simultaneously. Histogram is available to show the
probability distribution of a parameter. Trend is available to show
the parameter value vs. time

Advanced Math Function

In addition to the traditional (+, -, X, /) operations, FFT,
integration, differential, square root and so on are supported.
Formula Editor is available for more complex operations. 2 mathtraces are available

4

Hardware accelerated FFT supports up to 2 Mpts operation.
This provides high frequency resolution with a fast refresh rate.
The FFT function also supports a variety of window functions sothat it can adapt to different spectrum measurement needs.
Three modes (Normal, Average and Max hold) can satisfy
different requirements for observing the power spectrum. Autopeak detection and markers are supported

INT.SIGLENT.COM

SDS5000X Series Digital Storage Oscilloscope

Multiple Trigger Functions

Trigger Zone

Edge, Slope, Pulse, Video, Windows, Runt, Interval, Dropout,
Pattern, Qualified, Nth edge, Setup/hold, Delay and serial
trigger

Trigger Zone is available for advanced triggering

History Mode

Sequence Mode

History function can record up to 100,000 frames of waveforms.
The recording is executed automatically, so that the customer
can play back the history waveforms at any time in order toobserve unusual events and quickly locate the area of interest
using the cursors or measurements. The failed frames of Mask
Test can be stored as history

Segmented memory collection will store the waveform into
multiple memory segments (up to 100,000) and each segment
will store a triggered waveform as well the dead time
information. The interval between segments can be as small as
2 μs. All of the segments can be played back using the History
function

Search and Navigate

Serial Bus Decode

The SDS5000X can search events specified by the user in aframe. Events flagged by the Search can be recalled
automatically using Navigate. It can also navigate by time (delay
position) and history frames

Display the decoded characters through the events list. Bus
protocol information can be quickly and intuitively displayed intabular form. I2C, SPI, UART, CAN, LIN, CAN FD, FlexRay, I2S,
MIL-STD-1553B, SENT, Manchester and ARINC 429 aresupported

INT.SIGLENT.COM

5

SDS5000X Series Digital Storage Oscilloscope

Hardware-based Average and ERES Acquisition

NormalAverage

Waveform Histogram

ERES

Average and ERES (Enhanced Resolution) acquisition modes
are hardware-based, allowing the waveforms to be captured ata faster rate

The Waveform Histogram feature provides a statistics view ofthe waveform in horizontal and vertical directions

Hardware-based High Speed Mask Test Function

The SDS5000X utilizes a hardware-based Mask Test function,
performing up to 18,000 Pass / Fail decisions each second. It iseasy to generate user-defined test templates in order to provide
trace mask comparisons, making it suitable for long-term signal
monitoring or automated production line testing

6

Built-in Mask Editor application helps to create custom masks

INT.SIGLENT.COM

SDS5000X Series Digital Storage Oscilloscope

Bode PlotPower Analysis (Optional)

The SDS5000X can control the USB AWG module or a standalone SIGLENT SDG generator, to scan the amplitude andphase frequency response of the DUT, and display the data asa Bode Plot. This makes it possible to replace expensive
network analyzer in some applications

The Power Analysis option provides a full suite of power
measurements and analysis, which greatly improve themeasurement efficiency in switching power supplies and powerdevices design

Digital Channels / MSO

Four analog channels plus 16 digital channels enable users toacquire and trigger on the waveforms then analyze the pattern,simultaneously with one instrument

INT.SIGLENT.COM

25 MHz Function/Arbitrary Waveform Generator(Optional)

The SDS5000X can control the SAG1021I USB
Function/Arbitrary waveform generator to output waveform withup to 25 MHz frequency and ±3 V amplitude. Six basic
waveforms plus multiple types of arbitrary waveforms are builtin

7

SDS5000X Series Digital Storage Oscilloscope

8Complete Connectivity

Web Control

USB 2.0 Host, USB 2.0 Device (USBTMC), LAN 10M/100M(LXI, VXI-11, telnet, socket, web), Pass/ Fail, Trigger Out, 10
MHz In/Out and VGA output

With the embedded web server, users can control theoscilloscope from a simple web page. This provides wonderful
remote troubleshooting and monitoring capabilities

INT.SIGLENT.COM

SDS5000X Series Digital Storage Oscilloscope

Specifications
All specifications are not guaranteed unless the following conditions are met:
• The oscilloscope calibration period is current• The oscilloscope has been working continuously for at least 30 minutes at the specified temperature (18℃ ~ 28℃ )Acquire System (analog channel)
Sample rate
Memory depth
Peak detect
Average
ERES
Memorymanagement
mode
Sequence
History
Interpolation

5 GSa/s (interleaving mode), 2.5 GSa/s (non-interleaving mode)
250 Mpts (interleaving mode), 125 Mpts (non-interleaving mode)
400 ps4, 16, 32, 64, 128, 256, 512, 1024, 2048, 4096, 8192, 16384, 32768, 65536
Enhanced bit: 0.5, 1, 1.5, 2, 2.5, 3Auto, Fixed Sample Rate, Fixed Memory Depth
Up to 100,000 segments, interval between triggers = 2 μs min
Up to 100,000 frames
sinx/x, x

Vertical System(analog channel)
Bandwidth (-3dB) @50 Ω
Rise time (typical) @50 Ω
Bandwidth (-3dB) @1 MΩ,with probe
Vertical range
Vertical scale (probe 1X)
DC gain accuracy
Offset accuracy

Offset range (probe 1X)

Bandwidth flatness (>2
mV/div, @50 Ω)
Bandwidth limit
Low frequency response(AC coupling -3 dB)
Overshoot (150 ps pulse@50 Ω)
Coupling
Impedance
Max. Input voltage
SFDR
CH to CH Isolation (@50Ω)Probe Attenuation

SDS5034X

SDS5054X

SDS5104X

350 MHz*1.0 ns500 MHz**0.7 ns1 GHz**
0.4 ns350 MHz*

500 MHz**

500 MHz**

8 divisions
1 MΩ: 500 μV/div – 10 V/div(setting range), 1 mV/div – 10 V/div(specified range)
50 Ω: 500 μV/div – 1 V/div(setting range), 1 mV/div – 1 V/div(specified range)< 1.5%, ≥5mV/div< 3.0%, <5mV/div±(1.5%*offset+1.5%*full scale+1mV)
0.5mV/div~20mV/div: ±2V;***20.5mV/div~100mV/div: ±5V;
0.5mV/div~100mV/div: ±2V;
102mV/div~200mV/div: ±20V;
102mV/div~1V/div: ±20V;
205mV/div~1V/div: ±50V;
1.02V/div~10V/div: ±200V
1.02V/div~2V/div: ±200V
2.05V/div~10V/div: ±400V
50 kHz ~ BW/10: ±0.5 dB
BW/10 ~ BW/3: ±0.8 dB
BW/3 ~ BW2/3: +1.0 dB, -1.2 dB
BW2/3 ~ BW: +2.0 dB, -2.5 dB
20 MHz (±40%)
200 MHz (±40%)
5 Hz (typical)<10% (typical)<10% (typical)<15% (typical)

DC, AC, GND
DC1M: (1 MΩ±2%) || (16 pF±2 pF)
AC1M: (1.2 MΩ±2%) || (16 pF±2 pF)
50 Ω: 50 Ω±1%
1MΩ ≤ 400Vpk(DC + AC), DC~10kHz
50Ω ≤ 5Vrms, ±10V Peak≥ 32 dBc
DC ~ 100 MHz >40 dB
100 MHz ~ BW: ≥34 dB
1X, 10X, 100X, custom* Below 1 mV/div (included) the bandwidth is limited to 200 MHz(±40%)** Below 2.45 mV/div (included) the bandwidth is limited to 200 MHz(±40%)*** The SDS5054X upgraded from SDS5034X follows the Offset range spec of the SDS5034X

Horizontal System

SDS5034X

SDS5054X

SDS5104X

Time scale

1 ns/div – 1000 s/div

500 ps/div – 1000 s/div

200 ps/div – 1000 s/div

INT.SIGLENT.COM

9

SDS5000X Series Digital Storage Oscilloscope
Waveform update rate
Intensity grading
Display mode
Roll mode
Skew (C1~C4)
Time base Accuracy

Up to 110,000 wfm/s
256-level
Y-T, X-Y, Roll
≥ 50 ms/div< 150 ps
±1ppm initial; ±1ppm 1st year aging; ±3.5ppm 10-year aging

Trigger System
Mode

Auto, Normal, Single
Internal: ±4.1 div from the center of the screen
EXT: ±0.61 V
EXT/5: ±3.05 V

Level
Ext Trigger
input voltage

ChannelHold off range

Coupling

Accuracy (typical)

1 MΩ: ≤ 42 Vpk
50 Ω: ≤ 1.5 Vrms @ EXT, 5 Vrms @ EXT/5
By time: 8 ns ~ 30 s (8 ns step)
By event: 1 ~ 108
C1~C4
DC: Passes all components of the signal
AC: Blocks DC components and attenuates signals below 8 Hz
LFRJ: Attenuates the frequency components below 1.2 MHz
HFRJ: Attenuates the frequency components above 740 kHz
Noise RJ: Increases the trigger hysteresis
EXT
DC: Passes all components of the signal
AC: Blocks DC components and attenuates signals below 10 Hz
LFRJ: Attenuates the frequency components below 400 kHz
HFRJ: Attenuates the frequency components above 1.6 MHz
C1～C4: ±0.2div
EXT: ±0.3div
C1 ～ C4:

Sensitivity

EXT:

EXT/5:

Jitter
Displacement
Zone
Edge Trigger
SourceSlope
Slope Trigger
SourceSlope
Limit range>10mV/div:
5mV/div~10mV/div:≤ 2mV/div:

Noise RJ = ON
±0.35div±0.35 div
±0.75 div

200mVpp, DC ～ 10MHz
300mVpp, 10MHz ～ bandwidth (300 MHz typical)
1Vpp, DC ～ 10MHz
1.5Vpp, 10MHz ～ bandwidth (300 MHz typical)<9ps RMS (typical) for ≥300MHz sine and ≥6 divisions peak to peak amplitude for vertical gain
settings from 2.5mV/div to 10V/div<5ps RMS (typical) for ≥500MHz sine and ≥6 divisions peak to peak amplitude for vertical gain
settings from 2.5mV/div to 10V/div
Pre-Trigger: 0 ~ 100% memory
Delay-Trigger: 0 ~ 10,000 div
Up to 2 zones
Source: C1~C4
Property: Intersect, Not Intersect
C1~C4/EXT/(EXT/5)/AC Line/D0~D15
Rising, Falling, Rising & Falling
C1~C4
Rising, Falling<, >, in range, out of range

Time range

2ns ～ 20s，Resolution = 1 ns

Pulse Width Trigger
SourcePolarity
Limit range

C1~C4/D0~D15+wid, -wid<, >, in range, out of range

Time range

2ns ～ 20s，Resolution = 1 ns

Video Trigger
SourceC1~C4

10

Noise RJ = OFF±0.15div±0.25 div±0.5 div

INT.SIGLENT.COM

SDS5000X Series Digital Storage OscilloscopeStandard

NTSC，PAL，720p/50，720p/60，1080p/50，1080p/60，1080i/50，1080i/60，Custom

Synchronization
Trigger Condition
Window Trigger
SourceWindow typeInterval Trigger
SourceSlope
Limit range

Any, SelectLine, Field

Time range

2ns ～ 20s，Resolution = 1 ns

Dropout Trigger
SourceTimeout type
Slope

C1~C4/D0~D15
Edge, State
Rising, Falling

Time range

2ns ～ 20s，Resolution = 1 ns

Runt Trigger
SourcePolarity
Limit range

C1~C4
Positive, Negative<, >, in range, out of range

Time range

2ns ～ 20s，Resolution = 1 ns

Pattern Trigger
SourcePattern Setting
Logic
Limit range

C1~C4/D0~D15
Don’t Care, Low, HighAND, OR, NAND, NOR<, >, in range, out of range

Time range

2ns ～ 20s，Resolution = 1 ns

Qualified Trigger
Type
Qualified Source
Edge Trigger Source
Nth Edge Trigger
SourceSlope
Idle time

C1~C4
Absolute, Relative
C1~C4/D0~D15
Rising, Falling<, >, in range, out of range

State, State with Delay, Edge, Edge with Delay
C1~C4/D0~D15
C1~C4/D0~D15
C1~C4/D0~D15
Rising, Falling
8ns ～ 20s，Resolution = 1 ns

Edge Number

1 ～ 65535

Delay Trigger
Source A
Source B
Slope
Limit range

C1~C4/D0~D15
C1~C4/D0~D15
Rising, Falling<, >, in range, out of range

Time range

2ns ～ 20s，Resolution = 1 ns

Serial Trigger
SourceC1~C4/D0~D15

ProtocolI2 C

Standard：I2C、SPI、UART、CAN、LIN
Optional：CAN FD、FlexRay、I2S、MIL-STD-1553B、SENT

SPI
UART
CAN
LIN

Type: Start, Stop, Restart, No Ack, EEPROM, Address & Data, Data Length
Type: Data
Type: Start, Stop, Data, Parity Error
Type: All, Remote, ID, ID+Data, Error
Type: Break, Frame ID, ID+Data, Error

CAN FD（Optional）

Type: Start, Remote, ID, ID+Data, Error

FlexRay（Optional）

Type: TSS, Frame, Symbol, Errors

I2S（Optional）

Type: Data, Mute, Clip, Glitch, Rising Edge, Falling Edge

MIL-STD-1553B（Optional）

Type：Transfer, Word, Error, Timing

INT.SIGLENT.COM

11

SDS5000X Series Digital Storage Oscilloscope
SENT（Optional）

Type：Start, Slow channel, Fast channel, Error

Serial Decoder
Decoders

2

Threshold-4.1 ～ 4.1 div

List

1 ～ 7 lines

Decoder type
I2 C
SourceSignalFull duplex

Address

7bit，10bit

SPI
SourceSignal

C1~C4/D0~D15

Edge SelectChip selectBit Order

C1~C4/D0~D15
SCL，SDA

CLK，MISO，MOSI，CS
Rising, Falling
Active high, Active low, Clock timeout
LSB，MSB

UART
SourceSignal

C1~C4/D0~D15

Data Width

5 bit，6 bit，7 bit，8 bit

Parity Check
Stop Bit

None, Odd, Even, Mark, Space

Idle Level
Bit Order

Low, HighCAN
SourceLIN
LIN Version
SourceBaud Rate

RX，TX

1 bit，1.5 bit，2 bit
LSB，MSB
C1~C4/D0~D15
Ver1.3，Ver2.0
C1~C4/D0~D15
600bps, 1200bps, 2400bps, 4800bps, 9600bps, 19200bps, Custom

CAN FD（Optional）
SourceNominal Baud Rate
Data Baud Rate

C1~C4/D0~D15
10 kbps, 25 kbps, 50 kbps, 100 kbps, 250 kbps, 1 Mbps, Custom
500 kbps, 1 Mbps, 2 Mbps, 5 Mbps, 8 Mbps, 10 Mbps, Custom

FlexRay（Optional）
SourceBaud Rate

C1~C4/D0~D15
2.5 Mbps, 5 Mbps, 10 Mbps, Custom

I2S（Optional）
SourceSignalAudio Variant
Start Bits
Data Bits

C1~C4/D0~D15
BCLK, WS, DATA
Audio-I2S, Audio-LJ, Audio-RJ
0~31
1~32

MIL-STD-1553B（Optional）
SourceC1~C4

SENT（Optional）
SourceC1~C4/D0~D15

Manchester（Optional）
SourceBaud Rate

C1~C4
500 bps~5 Mbps

ARINC429（Optional）
SourceBaud Rate
12C1~C4
12.5 kbps~100 kbps, tolerance 1%~20%
INT.SIGLENT.COM

SDS5000X Series Digital Storage Oscilloscope
Word format

L/SDI/D/SSM，L/D/SSM，L/D

Measurement
Automatic Measurement
SourceC1~C4, D0~D15, F1~F2, Ref, History, Z1~Z4
Mode
Simple, Advanced
Range
Screen, Gating
Custom Threshold
Upper, Middle, Lower
No. of Measurements
Display 12 measurements at the same time (Display mode = M2)
Max, Min, Pk-Pk, Top, Base, Amplitude, Mean, Cycle Mean, Stdev, Cycle Stdev, RMS, Cycle RMS,
Vertical Parameters
Median, Cycle Median, FOV, FPRE, ROV, RPRE, Level@Trigger
Period, Frequency, Time@max, Time@min, +Width, -Width, 10-90%Rise time, 90-10%Fall time, Rise
Horizontal Parameters
time, Fall time, +Burst Width, -Burst Width, +Duty Cycle, -Duty Cycle, Delay, Time@Middle, CycleCycle jitter+Area@DC, -Area@DC, Area@DC, Absolute Area@DC, +Area@AC, -Area@AC, Area@AC, Absolute
Miscellaneous
Area@AC, Cycles, Rising Edges, Falling Edges, Edges, Positive pulses, Negative pulses, Positive
Parametersslope, Negative slope
Delay Parameters
Phase, FRFR, FRFF, FFFR, FFFF, FRLR, FRLF, FFLR, FFLF, Skew, Tsu@R, Tsu@F, Th@R, Th@F
Statistics
Current, Mean, Min, Max, Sdev, Count, Histogram, Trend, Track
Statistics Count
Unlimited, 1~1024
Statistics Count in a
Up to 25,000
frame (AIM count)
Cursors
SourceC1~C4, D0~D15, F1~F2, Ref, Histogram
Manual : Time X1, X2, (X1-X2), (1/ΔT); Vertical Y1, Y2, (Y1-Y2)
Type
Track: Time X1, X2, (X1-X2)
MeasureMath
Trace
SourceOperation

FFT

F1, F2
C1~C4, F1~F2, Z1~Z4
FFT, +, -, x, ÷, ∫dt, d/dt, √, Identity, Negation, |x|, Sign, ex, 10x, ln, lg, Interpolation, Formula Editor
Length: 2 Mpts, 1 Mpts, 512 kpts, 256 kpts, 128 kpts, 64 kpts, 32 kpts, 16 kpts, 8 kpts, 4 kpts, 2 kpts
Window: Rectangular, Blackman, Hanning, Hamming, Flattop
Display: Full Screen, Split, Exclusive
Mode: Normal, Max hold, Average
Tools: Peaks, Markers

Analysis
SearchSourceMode
Copy setting
Navigate
Type
Mask Test

C1~C4, History
Edge, Slope, Pulse, Interval, Runt
Copy from trigger, Copy to trigger
Search event, Time, History frame

SourceC1~C4，Z1~Z4

Mask creating
Mask test speed
DVM
SourceMode
Plot
Bode Plot
SourceSupported signal
sources
Sweep type

Auto (Create mask), Customized (Mask Editor)
Up to 18,000 frames/s
C1~C4
DC mean, DC RMS, AC RMS, Peak-peak, Amplitude
Bar, Histogram, Trend

C1~C4
SAG1021I (Connection: USB),
SDG series waveform generators (Connection: USB, LAN)
Simple, Vari-level
Mode: Linear, Logarithmic
Frequency
Range: 10 Hz ~ 120 MHz
MeasureUpper cutoff frequency, Lower cutoff frequency, Bandwidth, Gain margin, Phase margin
Power Analysis (Optional)
INT.SIGLENT.COM

13

SDS5000X Series Digital Storage Oscilloscope
MeasureHistogram
SourceType
Counter
SourceFrequency resolution
Totalizer

Power quality, Current Harmonics, Inrush current, Switching loss, Slew rate, Modulation, Output
ripple, Turn on/turn off, Transient response, PSRR, Efficiency, SOA
C1~C4
Horizontal, Vertical, Both
C1~C4
7 digits
Counter on edges, supports Gate and Trigger

Digital Channels
No. of Channels
Max. Sampling Rate
Memory Depth
Min. Detectable Pulse
Width
Level Group
Level Range
Logic TypeSkew

16
1.25 GSa/s
62.5 Mpts/ch
3.3 ns
D0~D7, D8~D15-10V~10V
TTL, CMOS, LVCMOS3.3, LVCMOS2.5, Custom
D0~D15: ±1 sampling interval
Digital to Analog: ± (1 sampling interval +1 ns)

SAG1021I Waveform Generator (optional)
ChannelsMax. Output Frequency
Sampling Rate
Frequency Resolution
Frequency Accuracy
Vertical Resolution
Amplitude Range
Waveforms
Output ImpedanceProtection
Insulation Voltage
Sine
Frequency
Offset accuracy (10kHz)
Amplitude flatness
SFDR
Harmonic distortion1
25 MHz
125 MSa/s
1 μHz±50 ppm
14 bit-1.5 V ~ +1.5 V (into 50Ω)
-3 V ~ +3 V (into High-Z)
Sine, Square, Ramp, Pulse, DC, Noise, 45 Arbitrary
50 Ω ±2%Over voltage protection, Current limit±42 Vpk
1 μHz ～ 25 MHz±(1%*offset setting value +3 mVpp)±0.3 dB, compare to 10 kHz, 2.5 Vpp into 50Ω
DC ~ 1 MHz-60 dBc
1 MHz ~ 5 MHz -55 dBc
5 MHz ~ 25 MHz -50 dBc
DC ~ 5 MHz-50 dBc
5 MHz ~ 25 MHz-45 dBc

Square/Pulse
Frequency

1 μHz ～ 10 MHz

Duty cycle

1% ～ 99%

Edge< 24 ns (10% ～ 90%)

Overshoot< 3% (typical，1 kHz, 1 Vpp)

Pulse width
Jitter (cycle-cycle)
Ramp
Frequency> 50 ns< 500 ps + 10 ppm

Linearity
Channels< 0.1% of Pk-Pk (typical, 1 kHz, 1 Vpp, 50% symmetry)

1 μHz ～ 300 kHz
0% ～ 100%

DC
Offset range
Accuracy14±1.5 V (into 50 Ω)±3 V (into Hi-Z)±(|setting value|*1%+3 mV)
INT.SIGLENT.COM

SDS5000X Series Digital Storage Oscilloscope
Noise
Bandwidth (-3 dB)
Arb
Frequency>25 MHz
1 μHz ～ 5 MHz

Waveform memory
Sample rate
Wave import

16 kpts
125 MSa/s
From EasyWaveX, from U-disk, directly from waveform data of analog channels

I/OStandard
Pass/Fail
Calibration signal
passive probe

3 USB 2.0 Hosts, 1 USB 2.0 Device, 10M/100M LAN, AUX (Pass/Fail+Trigger Out), 10 MHz In/ Out
3.3 V TTL output
for

1 kHz, 3.3 V Square

Display
Display Type
Resolution
Contrast
Backlight

10.1”TFT LCD with capacitive touch screen
1024×600
500:1 typical
500 nit typical

Display Setting
Range
Display Type
Persistence Time
Color Display
Language
Built-in Help System

8 x 10 gridDot, vector
OFF, 1 s, 5 s, 10 s, 30 s, infinite
Normal, Color; Supports customer trace color
Simplified Chinese, Traditional Chinese, English, French, Japanese, German, Spanish, Russian,
Italian, Portuguese
Simplified Chinese, English

Environmental
Temperature
Humidity
AltitudeOperating: 0 °C ~ 40 °C
Non-operating: -20 °C ~ 60 °C
Operating: 85% RH, 40 °C , 24 hours
Non-operating: 85% RH, 65 °C, 24 hours
Operating: ≤3,000 m
Non-operating: ≤15,000 m
Meets EMC directive (2014/30/EU), meets or exceeds IEC 61326-1:2012/EN61326-1:2013 (Basic)
Conducted disturbance

CISPR 11/EN 55011

CLASS A group 1，150kHz30MHz

Radiated disturbance

CISPR 11/EN 55011

CLASS A group 1， 30MHz1GHz

Electrostatic discharge (ESD)

IEC 61000-4-2/EN 61000-4-2

4.0 kV（Contact），8.0 kV（Air）
10 V/m（80 MHz to 1 GHz）;

Radio-frequency
electromagnetic field Immunity
Electromagnetic
Compatibility

SafetyINT.SIGLENT.COM

IEC 61000-4-3/EN 61000-4-3

3 V/m（1.4 GHz to 2 GHz）;
1 V/m（2.0 GHz to 2.7GHz）

Electrical fast transients (EFT)IEC 61000-4-4/EN 61000-4-4

SurgesIEC 61000-4-5/EN 61000-4-5

Radio-frequency continuous
conducted Immunity

IEC 61000-4-6/EN 61000-4-6

2kV （Input AC Power Ports）
1kV （Line to line）
2kV （Line to ground）
3 V, 0.15-80MHz

Voltage Dips：
0% UT during half cycle;
0% UT during 1 cycle;
Voltage dips and interruptions
IEC 61000-4-11/EN 61000-4-11
70% UT during 25/30 cycles
Voltage interruptions：0% UTduring 250/300 cycles
UL 61010-1:2012/R: 2018-11; CAN/CSA-C22.2 No. 61010-1:2012/A1:2018-11.
15

SDS5000X Series Digital Storage Oscilloscope
UL 61010-2-030:2018; CAN/CSA-C22.2 No. 61010-2-030:2018.

Power Supply
Input Voltage &Frequency
Power consumption

100 ～ 240 Vrms 50/60Hz
100 W max., 70 W typical, 4 W typical in standby mode

Mechanical
Dimensions
Weight16

Length x Width x Height = 370 mm×144 mm×231 mm
Net Weight: 4.0 kg
Gross Weight: 5.6 kg

INT.SIGLENT.COM

SDS5000X Series Digital Storage Oscilloscope

Ordering Information
Model

Description

SDS5104X
SDS5054X
SDS5034X

1 GHz, 4 CH, 5 GSa/s (Max.)
500 MHz, 4 CH, 5 GSa/s (Max.)
350 MHz, 4 CH, 5 GSa/s (Max.)

Standard Accessories

Quantity

USB cable
Quick start
Passive probe (SP3050A)
Certificate of calibration
Power cord

111/channel
1
1

Optional Accessories

Part No.

25 MHz isolated USB function/arbitrary waveform generator
16-channel logic probe
Power Analysis deskew fixture
STB3 demo signal source
1.5 GHz high-speed passive probe
1 GHz active probe
High voltage probe
High voltage differential probe

SAG1021I
SPL2016
DF2001A
STB3SP6150A
SAP1000
HPB4010
DPB1300/DPB4080/DPB5150/DPB5150A/DPB5700/DPB5700A
CPL5100/CP4020/CP4050/CP4070/CP4070A
CP6030/CP6030A/CP6150/CP6500
SCP5030/SCP5030A/SCP5150/SCP5150A
SAP4000P
BAG-S2Current probe
Power rail probe
Bag

Options

Part No.

350 MHz to 500 MHz bandwidth upgrade (4-ch model) * (software)
500 MHz to 1 GHz bandwidth upgrade (4-ch model) (software)Power Analysis (software)
I2S trigger & decode (software)
MIL-STD-1553B trigger & decode (software)
FlexRay trigger & decode (software)
CAN FD trigger & decode (software)
SENT trigger & decode (software)
Manchester decode (software)
ARINC429 decode (software)

SDS-5000X-4BW05
SDS-5000X-4BW10
SDS-5000X-PA
SDS-5000X-I2S
SDS-5000X-1553B
SDS-5000X-FlexRay
SDS-5000X-CANFD
SDS-5000X-SENT
SDS-5000X-Manch
SDS-5000X-ARINC* SDS5034X cannot be upgraded to SDS5104X

INT.SIGLENT.COM

17

SDS5000X Series Digital Storage Oscilloscope

About SIGLENTSIGLENT is an international high-tech company, concentrating on R&D, sales,
production and services of electronic test & measurement instruments.

SIGLENT first began developing digital oscilloscopes independently in 2002.
After more than a decade of continuous development, SIGLENT has extended
its product line to include digital oscilloscopes, isolated handheld
oscilloscopes,

function/arbitrary

waveform

generators,

RF/MWsignalgenerators, spectrum analyzers, vector network analyzers, digital multimeters,
DC power supplies, electronic loads and other general purpose test
instrumentation. Since its first oscilloscope was launched in 2005, SIGLENT
has become the fastest growing manufacturer of digital oscilloscopes. We
firmly believe that today SIGLENT is the best value in electronic test &measurement.

Headquarters:
SIGLENT Technologies Co., Ltd
Add: Bldg No.4 & No.5, Antongda Industrial

Zone, 3rd Liuxian Road, Bao'an District,
Shenzhen, 518101, China
Tel: + 86 755 3688 7876
Fax: + 86 755 3359 1582
Email: sales@siglent.com
Website: int.siglent.com

North America:
SIGLENT Technologies America, Inc
6557 Cochran Rd Solon, Ohio 44139
Tel: 440-398-5800
Toll Free: 877-515-5551

Fax: 440-399-1211
Email: info@siglentna.com
Website: www.siglentna.com

Follow us on
Facebook: SiglentTech

Europe:SIGLENT Technologies Germany GmbH
Add: Staetzlinger Str. 70
86165 Augsburg, Germany
Tel: +49(0)-821-666 0 111 0
Fax: +49(0)-821-666 0 111 22
Email: info-eu@siglent.com
Website: www.siglenteu.com

18

INT.SIGLENT.COM