SDS1000X-U
Digital Oscilloscope

Data Sheet
Rev. DS010AH _E01A

Sept. 2020

Key Features
100MHz bandwidth
Real-time sampling rate up to 1 GSa/s

SDS1104X-U

The Siglent SPO technology
 Waveform capture rates up to 100,000 wfm/s(normal mode) and 400,000 wfm/s (sequencemode) Supports 256-level intensity grading and color
temperature display modes Record length up to 14 Mpts
 Digital trigger system
Intelligent trigger: Edge, Slope, Pulse Width, Window,
Runt, Interval, Time out (Dropout), Pattern

Product Overview

Serial bus triggering and decoding (Standard), supports

SIGLENT’s SDS1000X-U Series Super Phosphor Oscilloscopes
is available in one bandwidth, 100 MHz. It has a maximum
sample rate of 1 GSa/s and a maximum record length of 14
Mpts. For ease-of-use, the most commonly used functions
can be accessed with its user-friendly front panel design.
The SDS1000X-U series employs the Siglent SPO (Super Phosphor Oscilloscope) technology that provides excellent
signal fidelity and performance. It comes with an innovative
digital trigger system with high sensitivity and low jitter, anda waveform capture rate of 400,000 frames/sec (sequencemode). The SDS1000X-U also employs a 256-level intensity
grading display function and a color temperature display
mode not found in other models in this class. SIGLENT’s
latest oscilloscope offering supports multiple powerfultriggering modes including serial bus triggering. Serial bus
decoding for IIC, SPI, UART, CAN, and LIN bus types areincluded. The X-U models also include History waveformrecording and sequential triggering that enable extended
waveform recording and analysis. Another powerful
addition is the new 128k point FFT math function that gives
the SDS1000X-U very high frequency resolution whenobserving signal spectra. The digital design includes a
hardware co-processor that delivers measurements quickly
and accurately without slowing acquisition and front-panel
response.

SDS1000X-U

also

supports

searchingandnavigating. The features and performance of SIGLENT’s new
SDS1000X-U cannot be matched anywhere else in this priceclass.

protocols IIC, SPI, UART, CAN, LIN
Video trigger, supports HDTV
10 types of one-button shortcuts, supports Auto Setup,
Default, Cursors, Measure, Roll, History, Display/Persist,
Clear Sweep, Zoom and Print
Segmented acquisition (Sequence) mode, divides the
maximum record length into multiple segments (up to
80,000), according to trigger conditions set by the user,
with a very small dead time segment to capture thequalifying event
History waveform record (History) function (maximum
recorded waveform length is 80,000 frames)
Automatic measurement function for 38 parameters aswell as Measurement Statistics, Zoom, Gating, Math,History and Reference functions
128k pts FFT, supports Peaks and Markers
Math and measurement functions use all sampled data
points (up to 14 Mpts)
Mathfunctions(FFT,

addition,subtraction,

multiplication, division, integration, differential, square
root)
Preset key can be customized for user settings or factory“defaults”
Security Erase mode
High Speed hardware-based Pass/ Fail function
Search and navigate
Large 7-inch TFT-LCD display with 800 * 480 resolution
Multiple interface types: USB Host, USB Device (USB TMC), LAN, Pass / Fail, Trigger Out
Supports SCPI remote control commands
VXI-11+SCPI, Telnet (Port 5024) +SCPI and Socket (PortWWW.SIGLENT.COM
15025) +SCPI programming over LAN
Supports Multi-language display and embedded online
help

SDS1000X-U Series Digital Oscilloscope

Models and Key Specifications
Model
Bandwidth

100 MHz

Sample rate (Max.)

1 GSa/s (One channel), 500 MSa/s(Two channels), 250 MSa/s(Four channels)

Channels4

Memory depth (Max.)

14 Mpts

Waveform capture rate (Max.)

100,000 wfm/s (normal mode), 400,000 wfm/s (sequence mode)

Trigger type

Edge, Slope, Pulse Width, Window, Runt, Interval, Dropout, Pattern, Video

Serial Trigger and decoder
(Std)

2

SDS1104X-U

IIC, SPI, UART, CAN, LIN

I/O

USB Host, USB Device, LAN, Pass/Fail, Trigger Out

Probe (Std)4 pcs passive probe PP510

Display

7-inch TFT-LCD (800x480)

WeightWithout package 2.6 kg; With package 3.8 kg

WWW.SIGLENT.COM

SDS1000X-U Series Digital Oscilloscope

Functions & Characteristics
7 Inch TFT-LCD Display and 10 One-button Menus

7 -inch TFT -LCD display with 800 * 480 resolution



Most commonly used functions are accessible using 10 different one-button operation keys: Auto Setup, Default,
Cursor, Measure, Roll, History, Persist, Clear Sweep, Zoom, Print

Record Length of up to 14 Mpts

Using hardware-based Zoom technologies and max record

With a waveform capture rate of up to 400,000 wfm/s

length of up to 14 Mpts, users are able to oversample to(sequence mode), the oscilloscope can easily capture thecapture for longer time periods at higher resolution andunusual or low-probability events.

use the zoom feature to see more details within eachsignal.

WWW.SIGLENT.COM
3

Waveform Capture Rate up to 400,000 wfm/s

SDS1000X-U Series Digital Oscilloscope

256-Level Intensity Grading and Color Temperature Display

SPO display technology provides fast refresh rates. The

The color temperature display is similar to the intensity-

resulting intensity-graded trace is brighter for events that

graded trace function, but the trace occurrence isoccur with more frequency and dims when the events occur

represented by different colors (color “temperature”) aswith less frequency.

opposed to changes in the intensity of one color. Red colors
represent events that occur more frequently, while blue isused to mark points that occur less frequently

Serial Bus Decoding Function (Standard)

History

Waveforms(History)

Mode

and

Segmented Acquisition (Sequence)

SDS1000X-U displays the decoding through the events list.

Playback the latest triggered events using the history

Bus protocol information can be quickly and intuitivelyfunction. Segmented memory collection will store trigger

displayed in a tabular format.

events into multiple (Up to 80,000) memory segments, eachsegment will store triggered waveforms and timestamp ofeach frame.

4

WWW.SIGLENT.COM

SDS1000X-U Series Digital Oscilloscope

True measurement to 14 M points

Gate and Zoom Measurement

SDS1004X-U can measure all sampled data points up to 14

Through Gate and Zoom measurement, the user canMpts. This ensures the accuracy of measurements while thespecify an arbitrary interval of waveform data analysis andmath co-processor decreases measurement time andstatistics. This helps avoid measurement errors that can beincreases ease-of-use.

caused by invalid or extraneous data, greatly enhancing
the measurements’ validity and flexibility.

128k points used to calculate the FFT

The math co-processor enables FFT analysis of incoming

The SDS1000X-U utilizes a hardware-based Pass/Failsignals using up to 128k samples per waveform. Thisfunction, performing up to 40,000 Pass / Fail decisions eachprovides high frequency resolution with a fast refresh rate.

second. Easily generate user defined test templates

The FFT function also supports a variety of window

provide trace mask comparison making it suitable for long-

functions so that it can adapt to different spectrum

term signal monitoring or automated production line

measurement needs. It also supports Peaks, Markers, atesting.

variety of FFT points.

WWW.SIGLENT.COM
5

Hardware-Based High-Speed Pass/FailSDS1000X-U Series Digital Oscilloscope

Customizable Default Key

Search and Navigate

The current parameters of the oscilloscope can be preset to

The SDS1000X-U can search events specified by the user in

Default Key through the Save menu.

a frame. It can also navigate by time (delay position) andhistorical frames.

Complete Connectivity

SDS1000X-U supports USB Host, USB Device (USB-TMC),
LAN (VXI-11), Pass/Fail and Trigger Out

6

WWW.SIGLENT.COM

SDS1000X-U Series Digital Oscilloscope

Specifications
Acquisition System
Sampling Rate (Max.)

1 GSa/s (One channel), 500 MSa/s(Two channels), 250 MSa/s(Four channels)

Memory Depth (Max.)

14 Mpts

Peak Detect

2 ns

Average

Averages: 4,16, 32, 64, 128, 256, 512, 1024

ERES

Enhance bits: 0.5, 1, 1.5, 2, 2.5, 3

WaveforminterpolationSin(x)/x, Linear

Input
Channels4

Coupling

DC, AC, GND

Impedance

DC: (1 MΩ±2%) || (11 pF ±2 pF)

Max. Input voltage

1 MΩ: ≤400 Vpk (DC + Peak AC <=10 kHz)CH to CH Isolation

DC-Max BW :>40 dB

Probe attenuation

1E-6X ~ 1E6X

Vertical System
Bandwidth (-3dB)

100 MHz

Vertical Resolution

8-bit

Vertical Scale (Probe
1X)

1 mV/div – 10 V/div (1-2-5 sequence)

Offset Range (Probe

1 mV- 200 mV: ± 2 V

1X)

206 mV- 10 V: ± 100 V

Bandwidth limit

20 MHz ± 40%
DC- 10% (BW): ± 1 dB

Bandwidth Flatness

10% - 50% (BW): ± 2 dB
50% - 100% (BW): + 2 dB/-3 dB

Low frequency
response≤2 Hz (at input BNC)(AC coupling -3 dB)
Noise
SFDR including
harmonics
DC Gain Accuracy
Offset Accuracy
Rise time
WWW.SIGLENT.COM
7

ST-DEV ≤ 0.2 division (<2 mV/div)
ST-DEV ≤ 0.1 division (≥ 2 mV/div)≥ 35 dB≤ ± 3.0%: 5 mV/div-10 V/div≤ ± 4.0% : ≤ 2 mV/div± (1%* Offset+1.5%*8*div+2 mV): ≥2 mV/div± (1%* Offset+1.5%*8*div+500 uV): 1 mv/div
Typical 3.5 ns

SDS1000X-U Series Digital Oscilloscope

Overshoot

(500

ps

Pulse)< 10%

Horizontal System
Timebase Scale

2 ns/div-100 s/div

Channel Skew

<100 psWaveform

CaptureRate

Up to 100,000 wfm/s (normal mode), 400,000 wfm/s (sequence mode)

Intensity grading

256 Levels

Display Format

Y -T, X -Y, Roll

Timebase Accuracy±25 ppm

Roll Mode

50 ms/div-100 s/div (1-2-5 sequence)

Trigger System
Mode

Auto, Normal, Single

Level

Internal: ±4.5 div from the center of the screen

Hold off range

80 ns- 1.5 s
ACDC

Coupling

LFRJ
HFRJ
Noise RJ
DC: Passes all components of the signal

Coupling

Frequency

Response

AC: Blocks DC components and attenuates signals below 8Hz
LFRJ: Blocks the DC component and attenuates the low-frequency
components below 2 MHz
HFRJ: Attenuates the high-frequency components above 1.2 MHz

Accuracy (typical)

Internal: ±0.2 div

Sensitivity

DC - Max BW 0.6 div

Jitter<100 ps

Displacement

Pre-Trigger: 0 - 100% Memory
Delay Trigger: 0 to 10,000 div

Edge Trigger
Slope

Rising, Falling, Rising & Falling

SourceAll channels/AC Line

Slope Trigger
Slope

Rising, Falling

Limit Range＜, ＞, ＜＞, ＞＜

SourceAll channels

Time Range

2ns- 4.2s

Resolution

1ns

Pulse Width Trigger

8

Polarity+wid , -wid

Limit Range＜, ＞, ＜＞, ＞＜
WWW.SIGLENT.COM

SDS1000X-U Series Digital Oscilloscope

SourceAll channels

Pulse Range

2 ns - 4.2s

Resolution

1 ns

Video Trigger
Signal Standard

NTSC，PAL，720p/50，720p/60，1080p/50，1080p/60，1080i/50，1080i/60，Custom

SourceAll channels

Sync

Any, SelectTrigger condition

Line, Field

Window Trigger
Window Type

Absolute, Relative

SourceAll channels

Interval Trigger
Slope

Rising, Falling

Limit Range＜, ＞, ＜＞, ＞＜

SourceAll channels

Time Range

2 ns - 4.2 s

Resolution

1 ns

Dropout Trigger
Timeout Type

Edge, State

SourceAll channels

Slope

Rising, Falling

Time Range

2 ns - 4.2 s

Resolution

1ns

Runt Trigger
Polarity+wid , -wid

Limit Range＜, ＞, ＜＞, ＞＜

SourceAll channels

Time Range

2 ns - 4.2 s

Resolution

1 ns

Pattern Trigger
Pattern Setting

Invalid, Low, High

Logic

AND, OR, NAND, NOR

SourceAll channels

Limit Range＜, ＞, ＜＞, ＞＜Time Range

2 ns - 4.2 s

Resolution

1 ns

Serial Trigger
I2C Trigger
Condition

Start, Stop, Restart, No Ack, EEPROM, 7-bits Address & Data, 10-bits Address & Data, Data Length

Source(SDA/SCL)

All channels

Data format

Hex

Limit Range

EEPROM: =, >, <Data Length

EEPROM: 1byte

WWW.SIGLENT.COM
9

SDS1000X-U Series Digital Oscilloscope

Addr & Data: 1-2byte
Data Length: 1-12byte
R/W bit

Addr & Data: Read, Write, Do not care

SPI Trigger
Condition

Data

Source(CS/CL/Data)

All channels

Data format

BinaryData Length

4-96-bit

Bit Value0, 1, X

Bit Order

LSB, MSBUART Trigger
Condition

Start, Stop, Data, Parity Error

Source(RX/TX)

All channels

Data format

Hex

Limit Range=, >, <

Data Length

1 byteData Width5, 6, 7, 8-bits

Parity Check

None, Odd, Even, Space, Mark

Stop Bit

1, 1.5, 2-bits

Idle Level

High, Low

Baud Rate(Selectable)

600/1200/2400/4800/960019200/38400/57600/115200 bit/s

Baud Rate (Custom)300-5000000 bit/s

CAN Trigger
Condition

Start, Remote, ID, ID + Data, Error

SourceAll channels

ID

STD (11-bits), EXT (29-bit)

Data Format

Hex

Data Length

1 -2 byte

Baud Rate

5k/10k/20k/50k/100k/125k/250k/500k/800k/1 Mbit/s

LIN Trigger
Condition

Break, Frame ID, ID+Data, Error

SourceAll channels

ID

1byte

Data Format

Hex

Data Length

1-2byteBaud Rate(Selectable)
Baud Rate (Custom)

600/1200/2400/4800/9600/19200 bit/s
300 bit/s -20 kbit/s

SearchEventEdge, Slope, Pulse, Interval, Runt

Event Number

Y-T: 700

10WWW.SIGLENT.COM

SDS1000X-U Series Digital Oscilloscope

ROLL: No limitation
Stop After ROLL: 700

Serial Decoder
Decoders

2

I2C
SignalSCL, SDA

Address

7, 10 bits

Threshold-4.5 - 4.5 div

List

1- 7 lines

SPI
SignalSCL,MISO, MOSI

Edge SelectRising, Falling

Idle Level

Low, HighBit Order

MSB, LSB

Threshold-4.5 - 4.5 div

List

1- 7 lines

UART
SignalRX, TX

Data Width

5，6，7，8 bits

Parity Check

None, Odd, Even, Space, Mark

Stop Bit

1, 1.5, 2 bits

Idle Level

Low, High

Threshold-4.5 - 4.5 div

List

1- 7 lines

CAN
SignalCAN_H, CAN_L

SourceCAN_H, CAN_L

Threshold-4.5 - 4.5 div

List

1- 7 lines

LIN
LIN

Specification

Ver1.3, Ver2.0

Package Revision
Threshold-4.5 - 4.5 div

List

1- 7 lines

Measurement
SourceNumber of
Measurements

All channels, All channels in Zoom, Math, All References, History
Display 4 measurements at the same time. 5 measurements displayed in statistics table.

Measurement Range

Screen or Gate region

Measurement

38Types

WWW.SIGLENT.COM
11

SDS1000X-U Series Digital Oscilloscope

Parameters
Max

Highest value in input waveform

Min

Lowest value in input waveform

Pk-Pk

Difference between maximum and minimum data values

Ampl

Difference between top and base in a bimodal signal, or between max
and min in an unimodal signal

VerticalTop

Value of most probable higher state in a bimodal waveform

Base

Value of most probable lower state in a bimodal waveform

Mean

Average of all data values

Cmean

Average of data values in the first cycle

Stdev

Standard deviation of all data values

Cstd

Standard deviation of all data values in the first cycle

VRMS

Root mean square of all data values

Crms

Root mean square of all data values in the first cycle

FOV

Overshoot after a falling edge;(base -min)/Amplitude

FPRE

Overshoot before a falling edge;(max -top)/Amplitude

ROV

Overshoot after a rising edge;(max -top)/Amplitude

RPRE

Overshoot before a rising edge;(base -min)/Amplitude

Level@X

the voltage value of the trigger point

PeriodTime between the middle threshold points of two consecutive, likepolarity edges

Freq

Reciprocal of period+Wid

Width measured at 50% level and positive slope-Wid

Width measured at 50% level and negative slope

Rise Time

Duration of rising edge from 10 -90%

Fall Time

Duration of falling edge from 90 -10%

Bwid

Time from the first rising edge to the last falling edge, or the first fallingedge to the last rising edge at the 50% crossing

Horizontal+DutTime difference between the 50% threshold of a rising edge to the 50%
threshold of the next falling edge of the pulse-DutTime difference between the 50% threshold of a falling edge to the 50%
threshold of the next rising edge of the pulse

DelayTime from the trigger to the first transition at the 50% crossing

Time@Level

Time from the trigger to each rising edge at the 50% crossing.
When Statistics is Off, it shows the time from the trigger to the last risingedge at the 50% crossing.
When Statistics is On, it shows the Current, Mean, Min, Max, Standard
Deviation of time from the trigger to each rising edge at the 50% crossing
in multiple frames (number = Count).

Delay

Phase

Phase difference between two edges

FRR

Time from the first rising edge of channel A to the following first risingedge of channel B

FRF
12WWW.SIGLENT.COM

Time from the first rising edge of channel A to the following first fallingSDS1000X-U Series Digital Oscilloscopeedge of channel B
FFR

Time from the first falling edge of channel A to the following first risingedge of channel B

FFF

Time from the first falling edge of channel A to the following first fallingedge of channel B

LRRTime from the first rising edge of channel A to the last rising edge ofchannel B

LRF

Time from the first rising edge of channel A to the last falling edge ofchannel B

LFRTime from the first falling edge of channel A to the last rising edge ofchannel B

LFF

Time from the first falling edge of channel A to the last falling edge ofchannel B

Skew

Time of source A edge minus time of nearest source B edge

Manual : Time X1, X2, (X1 -X2), (1/ΔT)
Cursors

Voltage Y1, Y2, (Y1 -Y2)
Track: Time X1, X2, (X1 -X2)

Statistics

Current, Mean, Min, Max, Stdev, Count

Counter

Hardware 6-digit counter（channels are selectable）

Math
Operation+,- , * , / , FFT , d/dt , ∫dt , √FFT window

Rectangular, Blackman, Hanning, Hamming, Flattop

FFT display

Full Screen, Split, Exclusive

I/O
Standard

USB Host, USB Device, LAN, Pass/Fail, Trigger Out

Pass/Fail

3.3V TTL Output

Display(Screen)
Display Type

7–inch TFT LCD

Display Resolution

800×480 pixels

Display Color

24-bitContrast(Typical)

500:1

Backlight

300 nitsRange

8 x 14 divisions

Display(Waveform)
Display Mode

Dot, Vector

Persist Time

Off, 1 Sec, 5 Sec, 10 Sec, 30 Sec, Infinite

Color Display

Normal, Color

Screen Saver

1 min, 5 min, 10 min, 30 min, 1 hour, Off

WWW.SIGLENT.COM
13

SDS1000X-U Series Digital Oscilloscope

Language

Simplified Chinese, Traditional Chinese, English, French, Japanese, Korean, German, Russian,
Italian, Portuguese

Environments
Temperature
Humidity

HeightOperating: 0℃ - +40℃
Non-operating: -20℃ - + 60℃
Operating: 85% RH, 40 ℃, 24 hours
Non-operating: 85% RH, 65 ℃, 24 hours
Operating: ≤ 3000 m
Non-operating: ≤ 15,000 m

Standards
Electromagnetic
compatibility

Meets EMC directive (2014/30/EU), meets or exceeds IEC 61326-1:2012/EN61326-1:2013 (Basic)
Conducted disturbance

CISPR 11/EN 55011

Radiated disturbance

CISPR 11/EN 55011

Electrostatic discharge (ESD)

IEC 61000-4-2/EN 61000-4-2

Radio-frequency
electromagnetic

CLASS A group 1 ， 150kHz30MHz
CLASS A group 1 ， 30MHz1GHz
4.0 kV（Contact），8.0 kV（Air）
10 V/m（80 MHz to 1 GHz）;

field

IEC 61000-4-3/EN 61000-4-3

Immunity

3 V/m（1.4 GHz to 2 GHz）;
1 V/m（2.0 GHz to 2.7GHz）

Electrical fast transients (EFT)IEC 61000-4-4/EN 61000-4-4

SurgesIEC 61000-4-5/EN 61000-4-5

Radio-frequency continuous
conducted Immunity

IEC 61000-4-6/EN 61000-4-6

2kV （Input AC Power Ports）
1kV （Line to line）
2kV （Line to ground）
3 V, 0.15-80MHz
Voltage Dips：
0% UT during 1 cycle;

Voltage

dipsinterruptions

andIEC 61000-4-11/EN 61000-4-

40% UT during 10/12 cycles;

11

70% UT during 25/30 cycles
Voltage interruptions：0% UTduring 250/300 cycles

SafetyUL 61010-1:2012/R: 2018-11; CAN/CSA-C22.2 No. 61010-1:2012/A1:2018-11.
UL 61010-2-030:2018; CAN/CSA-C22.2 No. 61010-2-030:2018.

Power Supply
Input Voltage
Power

14
WWW.SIGLENT.COM

100 ～ 240 Vrms 50/60Hz
100 ～ 120 Vrms 400Hz
50 W Max

SDS1000X-U Series Digital Oscilloscope

Mechanical
Length: 312 mm
Dimensions

Width: 132.6 mm
Height: 151 mm

WeightN.W: 2.6 kg; G.W: 3.8 kg

Probes and Accessories
Probe

Passive

PictureModel

Specifications &Description

PP510

Bandwidth: 100MHz, 1X/10X, 1M/10Mohm,300V/600V

Bandwidth: 100 KHz, Max. continuous current: 20Arms
CP4020Peak current: 60A
Switch Ratio: 50mV/A, 5mV/A, Accuracy: 50mV/A (0.4A 10Apk)±2%, 5mV/A(1A-60Apk) ±2%, 9V battery source
Bandwidth: 1MHz, Max. continuous current: 50Arms, Peak

CP4050current: 140A
Switch Ratio: 500mV/A, 50mV/A
Accuracy: 500mV/A (20mA -14ApK)±3%±20mA , 50mV/A(200mA -100ApK)±4%±200mA, 50mV/A (100A -140ApK) ±15%max, 9V battery
sourceCurrent Probe

Bandwidth: 150kHz, Max. continuous current: 70Arms,
CP4070Peak current: 200A
Switch Ratio: 50mV/A, 5mV/A, Accuracy: 50mV/A (0.4A 10ApK)±2% , 5mV/A(1A -200ApK) ±2%, 9V battery source
Bandwidth: 50 MHz, Max. continuous current: 30Arms,

CP5030Peak current: 50A
Switch Ratio: 100mV/A, 1V/A, Accuracy: 1V/A (±1%±1mA),
100mV/A (±1%±10mA), DC12V/1.2A power adapter
Bandwidth: 100 MHz, Max. continuous current: 30Arms,

CP5030A

Peak current: 50A
Switch Ratio: 100mV/A, 1V/A, Accuracy: 1V/A (±1%±1mA),
100mV/A (±1%±10mA), DC12V/1.2A power adapter

WWW.SIGLENT.COM
15

SDS1000X-U Series Digital Oscilloscope

Bandwidth: 12 MHz, Max. continuous current: 150Arms,
CP5150Peak current: 300A
Switch Ratio: 100mV/A, 10mV/A, Accuracy: 100mV/A(±1%±10mA), 10mV/A (±1%±100mA), DC12V/1.2A power
adapter
Bandwidth: 5 MHz, Max. continuous current: 500Arms,

CP5500Peak current: 750A
Switch Ratio: 100mV/A, 10mV/A, Accuracy: 100mV/A(±1%±10mA), 10mV/A(±1%±100mA), DC12V/1.2A power
adapter
Bandwidth: 50MHz, Differential Range: 800V (DC + Peak

DPB4080

AC),
100X/200X/500X/1000X, Accuracy: ±1%, DC 9V/1A power
adapter

DPB5150

Bandwidth: 70MHz, Differential Range: 1500V (DC + PeakAC),50X/500X
Accuracy: ±2%, DC 5V/1A USB adapter
Bandwidth: 100MHz, Differential Range: 1500V (DC + Peak

Differential

DPB5150A

AC),
50X/500X , Accuracy: ±2%

Probe

DC 5V/1A USB adapter
Bandwidth: 70MHz, Differential Range: 7000V (DC + Peak
DPB5700

AC),
100X/1000X , Accuracy: ±2%,
DC 5V/1A USB adapter
Bandwidth: 100MHz

DPB5700A

Differential Range: 7000V (DC + Peak AC),
100X/1000X
Accuracy: ±2%
DC 5V/1A USB adapter
Bandwidth: 40MHzHigh Voltage

HPB4010Differential Range: DC 10kV, AC (rms): 7kV (sine), AC (Vpp):
20kV (Pulse)
1000X
Accuracy: ≤3%
ProvidesIsolated front
end

ISFE

isolation

betweenstandard

oscilloscope

channels, isolation between
the measured signal and ground. Uses USB 5V power
supply, plug and play.
The maximum input voltage allowed is up to ± 600Vpk.

16
WWW.SIGLENT.COM

SDS1000X-U Series Digital Oscilloscope

Demo Board

STB-3 Test

Output signals including square, sine, AM, fast edge, pulse,

Board

PWM, I2C,
CAN, LIN etc. Used in teaching and demonstrations.

Rack Mount

WWW.SIGLENT.COM
17

SDS1X-E-RMK

The height is 4U.

SDS1000X-U Series Digital Oscilloscope

Ordering Information
Ordering information
Product NameSDS1104X-U 100MHz Four Channels
USB Cable -1
Quick Start -1

Standard Accessories

Passive Probe -4
Certification -1
Power Cord -1

Optional Accessories

Isolated Front EndISFESTB Demo Source

STB-3

High Voltage Probe

HPB4010CP4020/CP4050/CP4070/

Current Probes

CP4070A/CP5030/CP5030A/
CP5150/CP5500

Differential Probes
Rack Mount

18
WWW.SIGLENT.COM

DPB4080/DPB5150/DPB5150A/DPB5700/DPB5700A
SDS1X-E-RMK

SDS1000X-U Series Digital Oscilloscope

WWW.SIGLENT.COM
19