SDS1000X-E
Digital Oscilloscope

Data Sheet
Rev. 04G

Key Features
100 MHz, 200 MHz bandwidth models
Two channel series have one 1 GSa/s ADC, four channel series have two 1

SDS1104X-E
SDS1204X-E
SDS1202X-E

GSa/s ADCs. When all channels are enabled, each channel has a maximum
sample rate of 500 MSa/s. When a single channel per ADC is active, it has
sample rate of 1 GSa/s
The Siglent SPO technology◼

Waveform capture rates up to 100,000 wfm/s (normal mode) and 400,000
wfm/s (sequence mode)◼

Supports 256-level intensity grading and color temperature display
modes

◼

Record length up to 14 Mpts◼

Digital trigger system

Intelligent trigger: Edge, Slope, Pulse Width, Window, Runt, Interval, Time out(Dropout), Pattern

Product Overview

Serial bus triggering and decoding (Standard), supports protocols IIC, SPI,UART, CAN, LIN

SIGLENT’s SDS1000X-E Super Phosphor Oscilloscopes feature two channel and fourVideo trigger, supports HDTV

channel models. The two channel model is available with a 200 MHz analog bandwidth,

Low background noise with voltage scales from 500 μV/div to 10 V/diva single ADC with 1 GSa/s maximum sample rate, and a single memory module with 14

10 types of one-button shortcuts, supports Auto Setup, Default, Cursors,

Mpts of sample memory. The four channel scope is available in 100 and 200 MHz models

Measure, Roll, History, Display/Persist, Clear Sweep, Zoom and Print

and incorporates two 1 GSa/s ADCs and two 14 Mpts memory modules. When all

Segmented acquisition (Sequence) mode, divides the maximum record length

channels are enabled, each channel has sample rate of 500 MSa/s and a standard record

into multiple segments (up to 80,000), according to trigger conditions set bylength of 7 Mpts. When only a single channel per ADC is active, the maximum sample

the user, with a very small dead time segment to capture the qualifying event

rate is 1 GSa/s and the maximum record length is 14 Mpts. For ease-of-use, the most

History waveform record (History) function (maximum recorded waveformcommonly used functions can be accessed with its user-friendly front panel design.

length is 80,000 frames)

The SDS1000X-E series employs a new generation of SPO (Super- Phosphor Oscilloscope)

Automatic measurement function for 38 parameters as well as Measurement

technology that provides excellent signal fidelity and performance. The system noise is

Statistics, Zoom, Gating, Math, History and Reference functions

also lower than similar products in the industry. It comes with a minimum vertical input

1 Mpts FFT. Four channel series support Peaks and Markers

range of 500 uV/div, an innovative digital trigger system with high sensitivity and low

Math and measurement functions use all sampled data points (up to 14 Mpts)

jitter, and a waveform capture rate of 400,000 frames/sec (sequence mode). The

Math

SDS1000X-E also employs a 256-level intensity grading display function and a color

integration, differential, square root)

temperature display mode not found in other models in this class. SIGLENT’s latest

Preset key can be customized for user settings or factory “defaults”

oscilloscope offering supports multiple powerful triggering modes including serial bus

Security Erase mode

triggering. Serial bus decoding for IIC, SPI, UART, CAN, LIN bus types are included. The X-

High Speed hardware-based Pass/ Fail function

E models also include History waveform recording, and sequential triggering that enable

MSO, 16 digital channels (four channel series only, option)

extended waveform recording and analysis. Another powerful addition is the new

Bode plot, Measuring Power Supply Control Loop Response (four-channel

1 million points FFT math function that gives the SDS1000X-E very high frequency

series only)

resolution when observing signal spectra. The new digital design also includes a

Search and navigate (four-channel series only)

hardware co-processor that delivers measurements quickly and accurately without

USB AWG module (four channel series only, option)

slowing acquisition and front-panel response. The features and performance of

Web Browser based control (four channel series only)

SIGLENT’s new SDS1000X-E cannot be matched anywhere else in this price class.

Large 7-inch TFT-LCD display with 800 * 480 resolution

The four channel series support even more functions, including: searching andMultiple interface types: USB Host, USB Device (USB -TMC), LAN, Pass / Fail,navigating, on-screen Bode plot, 16 digital channels (Option), an external USB powered

Trigger Out

25 MHz AWG module (Option), and an embedded application that allows remote control

Supports SCPI remote control commands

via web browser.

functions(FFT,

addition,subtraction,

multiplication,

division,

VXI-11+SCPI, Telnet (Port 5024) +SCPI and Socket (Port 5025) +SCPI
programming over LAN
Supports Multi-language display and embedded online help
Supports Label(four channel series only)
Supports counter(four channel series only)
Supports

dataloggerincluding

SampleLoggerandMeasurement

Logger(four channel series only)
Supports NTP(Network Time Protocol) (four channel series only)

WWW.SIGLENT.COM
1

SDS1000X-E Series Digital Oscilloscope

Models and Key Specifications

Model
Bandwidth

SDS1204X-E

SDS1104X-E
100 MHzSDS1202X-E
200 MHzTwo channel series have a single 1 GSa/s ADC, four channel series have two 1 GSa/s ADCs. When all
Sample rate (Max.)

channels are enabled, each channel has a maximum sample rate of 500 MSa/s. When a single channel
per pair is active, that channel has sample rate of 1 GSa/s

Channels
Memory depth (Max.)

2+EXT (two channel series)
7 Mpts/CH (not interleave mode);
14 Mpts/CH (interleave mode)

Waveform capture rate (Max.)

100,000 wfm/s (normal mode), 400,000 wfm/s (sequence mode)

Trigger type

Edge, Slope, Pulse Width, Window, Runt, Interval, Dropout, Pattern, Video

Serial Trigger and decoder (Std)

IIC, SPI, UART, CAN, LIN

16 Digital Channels (four channel
series only, option)

Maximum waveform capture rate up to 1 GSa/s, Record length up to 14 Mpts/CH

USB AWG module (four channel

One channel, 25 MHz, sample rate of 125 MHz, wave length of 16 kpts,

series only, option)

isolated output (SAG1021I only)

Bode plot ( four channel series only)
Data Logger(four channel series
only)

Minimum start frequency of 10 Hz, minimum scan bandwith of 500 Hz, maximum scan bandwidth of 120
MHz (dependent on Oscilloscope and AWG bandwidth), 500 maximum scan frequency points
Sample Logger. The Max sample rate is 25kSa/s, the Min sample rate is 1Sa/s.
Measure Logger. The Max interval is 10 minutes, the Min interval is 0.1s. The Max number of measurementsthat can be logged is 4.

I/O

USB Host, USB Device, LAN, Pass/Fail, Trigger Out, Sbus (Siglent MSO)

Probe (Std)4 pcs passive probe PP510

Display

7-inch TFT-LCD (800x480)

Weight2

4 (four channel series)

4/2 pcs passive probe PP215

Four channel series: Without package 2.6 kg; With package 3.8 kg
Two channel series: Without package 2.5 kg; With package 3.5 kg

WWW.SIGLENT.COM

Functions & Characteristics
7 Inch TFT-LCD Display and 10 One-button Menus

Front panel of the four channel series

Front panel of the two channel series

7-inch TFT -LCD display with 800 * 480 resolution. Most commonly used functions are accessible using 10 different one-button operation keys:
Auto Setup, Default, Cursor, Measure, Roll, History, Persist, Clear Sweep, Zoom, Print.

When all channels are enabled, each channel has a maximum sample rate of 500 MSa/s. When a single channel perpair is active, that channel has sample rate of 1 GSa/s

The four channel series has two 1GSa/s ADC chips (channel 1 and 2 share one, channel 3 and 4 share another), so that each channel can achieve
sample rates up to 500 MSa/s and work on bandwidths of 200 MHz when all channels are enabled.

Record Length of up to 14 Mpts

Using hardware-based Zoom technologies and max record length ofWith a waveform capture rate of up to 400,000 wfm/s (sequence

up to 14 Mpts, users are able to oversample to capture for longer timemode), the oscilloscope can easily capture the unusual or low-

periods at higher resolution and use the zoom feature to see moreprobability events.

WWW.SIGLENT.COM
3

Waveform Capture Rate up to 400,000 wfm/s

SDS1000X-E Series Digital Oscilloscopedetails within each signal.

256-Level Intensity Grading and Color Temperature Display

SPO display technology provides fast refresh rates. The resulting

The color temperature display is similar to the intensity-graded trace

intensity-graded trace is brighter for events that occur with morefunction, but the trace occurrence is represented by different colors

frequency and dims when the events occur with less frequency.

(color “temperature”) as opposed to changes in the intensity of one
color. Red colors represent events that occur more frequently, whileblue is used to mark points that occur less frequently.

Serial Bus Decoding Function (Standard)

History Waveforms (History) Mode and Segmented
Acquisition (Sequence)

SDS1000X-E displays the decoding through the events list. Bus protocol

Playback the latest triggered events using the history function.

information can be quickly and intuitively displayed in a tabular format.

Segmented memory collection will store trigger events into multiple(Up to 80,000) memory segments, each segment will store triggered
waveforms and timestamp of each frame.

True measurement to 14 M points

4

Gate and Zoom Measurement

WWW.SIGLENT.COM

SDS1000X-E can measure all sampled data points up to 14 Mpts. This

Through Gate and Zoom measurement, the user can specify anensures the accuracy of measurements while the math co-processor

arbitrary interval of waveform data analysis and statistics. This helps

decreases measurement time and increases ease-of-use.

avoid measurement errors that can be caused by invalid or extraneous
data, greatly enhancing the measurements’ validity and flexibility.

1M points used to calculate the FFT

Hardware-Based High-Speed Pass/FailThe new math co-processor enables FFT analysis of incoming signals

The SDS1000X-E utilizes a hardware-based Pass/Fail function,using up to 1 M samples per waveform. This provides high frequency

performing up to 40,000 Pass / Fail decisions each second. Easily

resolution with a fast refresh rate. The FFT function also supports agenerate user defined test templates provide trace mask comparison

variety of window functions so that it can adapt to different spectrum

making it suitable for long-term signal monitoring or automatedmeasurement needs. Four-channel series support Peaks, Markers, a

production line testing.

variety of numbers.

Customizable Default Key

16 Digital Channels/MSO (four-channel series only,
option)The current parameters of the oscilloscope can be preset to Default Key

16 digital channels enables users to acquire and trigger on thethrough the Save menu.

waveforms then analyze the pattern, simultaneously with one
instrument.

WWW.SIGLENT.COM
5

SDS1000X-E Series Digital Oscilloscope

Search and Navigate (four-channel series only)

USB 25MHz AWG Module (four-channel series only,
option)The SDS1000X-E can search events specified by the user in a frame. It

The four channel series supports a USB 25 MHz function/arbitrary

can also navigate by time (delay position) and historical frames.

waveform generator that is operated from the USB host connection.
Functions include Sine, Square, Ramp, Pulse, Noise, DC and 45 built-in
waveforms. The arbitrary waveforms can be accessed and edited by the
SIGLENT EasyWave PC software.

Bode Plot (four-channel series only)

SDS1000X-E can control the USB AWG module or control an independent SIGLENT SDG instrument, scan a devices amplitude and phase frequency
response, and display the data as a Bode Plot. There is also a Vari-level Mode for accurately measuring Power Supply Control Loop Response(PSRR). It can also show the result lists, and export the data to a USB disk.

Label(four-channel series only)

Web control (four-channel series only)

When multi-channel measurement is used, corresponding labels areWith the new embedded web server, users can control the SDS1xx4X-E

set for different sources to facilitate observation and reading.

from

a

simple

web

page.

This

provideswonderfulremotetroubleshooting and monitoring capabilities. The web page has PC andmobile styles that include an embedded virtual control panel.

6

WWW.SIGLENT.COM

Sample Logger(four-channel series only)

Measurement Logger (four-channel series only)

The Sample Logger is the mode of logging the sampling points overThe measurement Logger is designed to collect measurement values

long time durations. . Data is logged into the internal flash or external

for a long time. The data is logged in memory. After logging is

U disk in real time. After logging is complete, the user can recall thecomplete, the data can be saved into the internal flash or external U

sampling points on the oscilloscope, or analyze the saved data on the

disk.

computer.

Complete Connectivity

Back panel of the four-channel series

Back panel of the two-channel series

SDS1000X-E supports USB Host, USB Device (USB -TMC), LAN (VXI -11), Pass/Fail and Trigger Out.

Specifications
Acquisition System
Sampling Rate (Max.)

1 GSa/s (single channel/pair), 500 MSa/s (two channels/pair)

Memory Depth (Max.)Max 14 Mpts/Ch (single channel/pair), 7 Mpts/Ch (two channels/pair)

Peak Detect

2 ns (Four-channel series)
4 ns (Two-channel series)

Average

Averages:4,16, 32,64,128,256,512,1024

ERES

Enhance bits:0.5, 1.5, 2, 2.5, 3

Waveform interpolationSin(x)/x, Linear

Input
Channels

WWW.SIGLENT.COM
7

4 (Four-channel series)
2+EXT (Two-channel series)

SDS1000X-E Series Digital Oscilloscope
Coupling
Impedance

DC, AC, GND
DC: (1 MΩ±2%) || (15 pF ±2 pF) (Four-channel series)
DC: (1 MΩ±2%) || (18 pF ±2 pF) (Two-channel series)

Max. Input voltage

1 MΩ: ≤400 Vpk (DC + Peak AC <=10 kHz)CH to CH Isolation

DC-Max BW :>40 dB

Probe attenuation

0.1X,0.2X,0.5X,1X,2X,5X,10X……1000X,2000X,5000X,10000X,Custom

Vertical System
Bandwidth (-3dB)
Vertical Resolution
Vertical Scale (Probe 1X)
Offset Range (Probe 1X)
Bandwidth limit

200 MHz (SDS1204X-E/SDS1202X-E)
100 MHz (SDS1104X-E)
8-bit
500 μV/div–10 V/div (1-2-5 sequence )500 μV- 118 mV: ± 2 V
120 mV- 1.18 V: ± 20 V
1.2 V- 10 V: ± 200 V
20 MHz ± 40%
DC- 10% (BW): ± 1 dB

Bandwidth Flatness

10%- 50% (BW): ± 2 dB
50%- 100% (BW): + 2 dB/-3 dB

Low frequency response(AC coupling -3 dB)≤2 Hz (at input BNC)
ST-DEV ≤0.5 division (<1 mV/div)

Noise

ST-DEV ≤0.2 division (<2 mV/div)
ST-DEV ≤0.1 division(≥ 2 mV/div)

SFDR including harmonics
DC Gain Accuracy
Offset Accuracy
Rise timeOvershoot (500 ps Pulse)≥ 35 dB≤±3.0%: 5 mV/div-10 V/div≤±4.0%: ≤2 mV/div±(1%* Offset+1.5%*8*div+2 mV): ≥2 mV/div±(1%* Offset+1.5%*8*div+500 uV): ≤1 mv/div
Typical 1.8 ns (SDS1204X-E/SDS1202X-E)
Typical 3.5 ns (SDS1104X-E)< 10%

Horizontal System
Timebase Scale

1.0 ns/div-100 s/div

Channel Skew

<100 psWaveform Capture Rate

Up to 100,000 wfm/s (normal mode), 400,000 wfm/s (sequence mode)

Intensity grading

256 Levels

Display Format

Y -T, X -Y, Roll

Timebase Accuracy±25 ppm

Roll Mode

50 ms/div-100 s/div (1-2-5 sequence)

Trigger System
Mode

Auto, Normal, Single
Internal: ±4.5 div from the center of the screen

Level

EXT: ±0.6 V (Two channel series)
EXT/5: ±3 V(Two channel series)

Hold off range

80 ns- 1.5 s
ACCoupling

DC
LFRJ
HFRJ

8

WWW.SIGLENT.COM

Noise RJ
DC: Passes all components of the signal
Coupling

Frequency

Response

AC: Blocks DC components and attenuates signals below 8Hz
LFRJ: Blocks the DC component and attenuates the low-frequency
components below 2 MHz
HFRJ: Attenuates the high-frequency components above 1.2 MHz
DC: Passes all components of the signal

Coupling

Frequency

Response(EXT, Two channel series)

AC: Blocks DC components and attenuates signals below 20 Hz
LFRJ: Blocks the DC component and attenuates the low-frequency
components below 7 kHz
HFRJ: Attenuates the high-frequency components above 160 kHz

Accuracy (typical)

Internal: ±0.2 div
EXT(Two channel series): ±0.4 div
DC - Max BW 0.6 div
EXT(Two channel series): 200 mVpp DC– 10 MHz

Sensitivity

300 mVpp 10 MHz - BW frequency
EXT/5 (Two-channel series): 1 Vpp DC – 10 MHz；
1.5 Vpp 10 MHz -BW frequency

Jitter
Displacement<100 ps
Pre-Trigger: 0 - 100% Memory
Delay Trigger: 0 to 10,000 div

Edge Trigger
Slope
SourceRising, Falling, Rising & Falling
All channels/EXT/(EXT/5)/AC Line(Two-channel series)
All channels/ AC Line(Four-channel series)

Slope Trigger
Slope

Rising, Falling

Limit Range＜, ＞, ＜＞, ＞＜

SourceAll channels

Time Range

2ns- 4.2s

Resolution

1ns

Pulse Width Trigger
Polarity+wid , -wid

Limit Range＜, ＞, ＜＞, ＞＜

SourceAll channels

Pulse Range

2 ns - 4.2s

Resolution

1 ns

Video Trigger
Signal Standard

NTSC, PAL,720p/50,720p/60,1080p/50,1080p/60,1080i/50,1080i/60,Custom

SourceAll channels

Sync

Any, SelectTrigger condition

Line, Field

Window Trigger
Window Type

Absolute, Relative

SourceAll channels

Interval Trigger
Slope

Rising, Falling

Limit Range＜, ＞, ＜＞, ＞＜

SourceAll channels

Time Range

2 ns - 4.2 s

Resolution

1 ns

WWW.SIGLENT.COM
9

SDS1000X-E Series Digital Oscilloscope

Dropout Trigger
Timeout Type

Edge, State

SourceAll channels

Slope

Rising, Falling

Time Range

2 ns - 4.2 s

Resolution

1ns

Runt Trigger
Polarity+wid , -wid

Limit Range＜, ＞, ＜＞, ＞＜

SourceAll channels

Time Range

2 ns - 4.2 s

Resolution

1 ns

Pattern Trigger
Pattern Setting

Invalid, Low, High

Logic

AND, OR, NAND, NOR

SourceAll channels

Limit Range＜, ＞, ＜＞, ＞＜Time Range

2 ns - 4.2 s

Resolution

1 ns

Serial Trigger
I2C Trigger
Condition

Start, Stop, Restart, No Ack, EEPROM, 7-bits Address & Data, 10-bits Address & Data, Data Length

Source(SDA/SCL)

All channels

Data format

Hex

Limit Range

EEPROM: =, >, <Data Length

EEPROM: 1byte
Addr & Data: 1-2byte
Data Length: 1-12byte

R/W bit

Addr & Data: Read, Write, Do not care

SPI Trigger
Condition

Data

Source(CS/CL/Data)

All channels

Data format

BinaryData Length

4-96-bit

Bit Value0, 1, X

Bit Order

LSB, MSBUART Trigger
Condition

Start, Stop, Data, Parity Error

Source(RX/TX)

All channels

Data format

Hex

Limit Range=, >, <

Data Length

1 byteData Width5, 6, 7, 8-bits

Parity Check

None, Odd, Even, Space, Mark

Stop Bit

1, 1.5, 2-bits

Idle Level

High, Low

Baud Rate(Selectable)

600/1200/2400/4800/960019200/38400/57600/115200/Custom bit/s

Baud Rate (Custom)300-5000000 bit/s

CAN Trigger
Condition

10
WWW.SIGLENT.COM

Start, Remote, ID, ID + Data, Error

Source

All channels

ID

STD (11-bits), EXT (29-bit)

Data Format

Hex

Data Length

1 -2 byte

Baud Rate

5k/10k/20k/50k/100k/125k/250k/500k/800k/1M/Custom bit/s

LIN Trigger
Condition

Break, Frame ID, ID+Data, Error

SourceAll channels

ID

1byte

Data Format

Hex

Data Length

1-2byteBaud Rate(Selectable)
Baud Rate (Custom)

600/1200/2400/4800/9600/19200/Custom bit/s
300 bit/s -20 kbit/s

SearchEventEdge, Slope, Pulse, Interval, Runt
Y-T: 700

Event Number

ROLL: No limitation
Stop After ROLL: 700

Serial Decoder
Decoders

2

IC
2

SignalSCL, SDA

Address

7, 10 bits

Threshold-4.5 - 4.5 div

List

1- 7 lines

SPI
SignalSCL,MISO, MOSI, CS (2 channel scopes can only use 2 signal) identifiers

Edge SelectRising, Falling

Idle Level

Low, HighBit Order

MSB, LSB

Threshold-4.5 - 4.5 div

List

1- 7 lines

UART
SignalRX, TX

Data Width

5，6，7，8 bits

Parity Check

None, Odd, Even, Space, Mark

Stop Bit

1, 1.5, 2 bits

Idle Level

Low, High

Threshold-4.5 - 4.5 div

List

1- 7 lines

CAN
SignalCAN_H, CAN_L

SourceCAN_H, CAN_L, CAN_H-CAN_L

Threshold-4.5 - 4.5 div

List

1- 7 lines

LIN
WWW.SIGLENT.COM
11

SDS1000X-E Series Digital Oscilloscope
LIN Specification Package

Ver1.3, Ver2.0

Revision
Threshold-4.5 - 4.5 div

List

1- 7 lines

Measurement
SourceNumber of
Measurements

All channels, All channels in Zoom, Math, All References, History
Display 4 measurements at the same time. 5 measurements displayed in statistics table.

Measurement Range

Screen or Gate region

Measurement Parameters

38 Types
Max

Highest value in input waveform

Min

Lowest value in input waveform

Pk-Pk

Difference between maximum and minimum data values

Ampl

Difference between top and base in a bimodal signal, or between max and min in an
unimodal signal

VerticalTop

Value of most probable higher state in a bimodal waveform

Base

Value of most probable lower state in a bimodal waveform

Mean

Average of all data values

Cmean

Average of data values in the first cycle

Stdev

Standard deviation of all data values

Cstd

Standard deviation of all data values in the first cycle

VRMS

Root mean square of all data values

Crms

Root mean square of all data values in the first cycle

FOV

Overshoot after a falling edge;(base -min)/Amplitude

FPRE

Overshoot before a falling edge;(max -top)/Amplitude

ROV

Overshoot after a rising edge;(max -top)/Amplitude

RPRE

Overshoot before a rising edge;(base -min)/Amplitude

Level@X

the voltage value of the trigger point

PeriodTime between the middle threshold points of two consecutive, like-polarity edges

Freq

Reciprocal of period+Wid

Width measured at 50% level and positive slope-Wid

Width measured at 50% level and negative slope

Rise Time

Duration of rising edge from 10 -90%

Fall Time

Duration of falling edge from 90 -10%

Bwid

Time from the first rising edge to the last falling edge, or the first falling edge to the lastrising edge at the 50% crossing+Dut
Horizontal

Time difference between the 50% threshold of a rising edge to the 50% threshold of thenext falling edge of the pulse-DutTime difference between the 50% threshold of a falling edge to the 50% threshold of thenext rising edge of the pulse

DelayTime from the trigger to the first transition at the 50% crossing

Time@Level

Time from the trigger to each rising edge at the 50% crossing.
When Statistics is Off, it shows the time from the trigger to the last rising edge at the 50%
crossing.
When Statistics is On, it shows the Mean, Min, Max, Standard Deviation of time from thetrigger to each rising edge at the 50% crossing in multiple frames (number = Count). The
Current shows the time of current frame from the trigger to the last rising edge at the 50%
crossing.

Delay

12
WWW.SIGLENT.COM

Phase

Phase difference between two edges

FRR

Time from the first rising edge of channel A to the following first rising edge of channel B

FRFTime from the first rising edge of channel A to the following first falling edge of channel
B

FFR

Time from the first falling edge of channel A to the following first rising edge of channel
B

FFF

Time from the first falling edge of channel A to the following first falling edge of channel
B

LRR

Time from the first rising edge of channel A to the last rising edge of channel B

LRF

Time from the first rising edge of channel A to the last falling edge of channel B

LFRTime from the first falling edge of channel A to the last rising edge of channel B

LFF

Time from the first falling edge of channel A to the last falling edge of channel B

Skew

Time of source A edge minus time of nearest source B edge

Manual : Time X1, X2, (X1 -X2), (1/ΔT)
Cursors

Voltage Y1, Y2, (Y1 -Y2)
Track: Time X1, X2, (X1 -X2)

Statistics

Current, Mean, Min, Max, Stdev, Count

Counter

Hardware 6-digit counter（channels are selectable）

Math
Operation+,

FFT window

Rectangular, Blackman, Hanning, Hamming, Flattop- , * , / , FFT , d/dt , ∫dt , √FFT display

Full Screen, Split, Exclusive

USB AWG Module (four channel series only, option)
Channel1Max. Output Frequency

25 MHzSampling Rate

125 MSa/s

Frequency Resolution

1 μHz

Frequency Accuracy±50 ppm

Vertical Resolution

14-bits

Amplitude Range-1.5 ～ +1.5 V (50Ω load)-3 ～ +3 V (High-Z load)

Waveform TypeSine, Square, Ramp, Pulse, Noise, DC and 45 built-in waveforms

Output impedance

50Ω ± 2%

Protection

Over-Voltage Protection, Current-Limiting Protection

Insulation Voltage±42 Vpk (for SAG2021I only)

Sine
Frequency

1 μHz～ 25 MHz

Offset Accuracy(10 kHz)±(1%*Offset Setting Value +3 mVpp)

Amplitude flatness±0.3 dB(10 kHz, 5Vpp)
SFDR

DC～1 MHz-60dBc1 MHz～5 MHz-55dBc5 MHz～25 MHz -50dBc
HD

DC～5 MHz-50dBc5 MHz～25 MHz -45dBc

Square/Pulse
Frequency

1 μHz～ 1 0MHz

Duty Cycle

1% ～99%Rise/Fall time< 24 ns (10% ～ 90%)

Overshoot(1kHz,1Vpp,< 3% (typical 1kHz, 1 Vpp)

WWW.SIGLENT.COM
13

SDS1000X-E Series Digital Oscilloscope
Typical)Pulse Width> 50 ns

Jitter< 500 ps + 10 ppm

Ram
Frequency

1 μHz～ 300 kHz

Linearity(Typical)< 0.1% of Pk-Pk (Typical, 1 kHz, 1 Vpp, 50% Symmetry)

Symmetry

0% ～ 100%

DC
Offset range±1.5 V (50Ω load)±3 V (High-Z load)

Accuracy±(|offset|*1%+3 mV)

NoiseBandwidth>25 MHz (-3dB)

Arbitrary Wave
Frequency

1 μHz～ 5 MHz

Wave Length

16 kpts

Sampling Rate

125MSa/s

Lead in

EasyWave and U-Disk

Digital Channels (four channel series only, option)
No. of Channels

16

Max. Sampling Rate

1 GSa/sMemory Depth
Min.

Detectable

14 Mpts/CH
Pulse

Width

4 ns

Level GroupD0~D7, D8~D15

Level Range-8 V ~ 8 V

Logic Type

TTL, CMOS, LVCMOS3.3, LVCMOS2.5, Custom

Skew

D0~D15: ±1 sampling interval
Digital to Analog: ± (1 sampling interval +1ns)

I/O
Standard

USB Host (1 for two channel series, and 2 for four channel series), USB Device, LAN, Pass/Fail, Trigger Out

Pass/Fail

3.3V TTL Output

Display(Screen)
Display Type

7–inch TFT LCD

Display Resolution

800×480 pixels

Display Color

24-bitContrast(Typical)

500:1

Backlight

300 nits

Display(Waveform)Range

8 x 14 divisions

Display Mode

Dot, Vector

Persist Time

Off, 1 Sec, 5 Sec, 10 Sec, 30 Sec, Infinite

Color Display

Normal, Color

Screen Saver

1 min, 5 min, 10 min, 30 min, 1 hour, Off

14
WWW.SIGLENT.COM

Language

Simplified Chinese, Traditional Chinese, English, French, Japanese, Korean, German, Russian, Italian, Portuguese

Environments
Temperature
Humidity

HeightOperating: 0℃ - +40℃
Non-operating: -20℃ - + 60℃
Operating: 85% RH, 40 ℃, 24 hours
Non-operating: 85% RH, 65 ℃, 24 hours
Operating: ≤ 3000 m
Non-operating: ≤ 15,000 m

Standards
Electromagnetic

Meets EMC directive (2014/30/EU), meets or exceeds IEC 61326-1:2012/EN61326-1:2013 (Basic)compatibility

Conducted disturbance

CISPR 11/EN 55011

CLASS A group 1，150kHz-30MHz

Radiated disturbance

CISPR 11/EN 55011

CLASS A group 1， 30MHz-1GHz

Electrostatic discharge (ESD)

IEC 61000-4-2/EN 61000-4-2

4.0 kV（Contact），8.0 kV（Air）

Radio-frequency

electromagnetic

field Immunity

10 V/m（80 MHz to 1 GHz）;
IEC 61000-4-3/EN 61000-4-3

1 V/m（2.0 GHz to 2.7GHz）

Electrical fast transients (EFT)IEC 61000-4-4/EN 61000-4-4

SurgesIEC 61000-4-5/EN 61000-4-5

Radio-frequency

3 V/m（1.4 GHz to 2 GHz）;

continuous

conducted Immunity

IEC 61000-4-6/EN 61000-4-6

2kV （Input AC Power Ports）
1kV （Line to line）
2kV （Line to ground）
3 V, 0.15-80MHz
Voltage Dips：
0% UT during 1 cycle;

Voltage dips and interruptions

IEC 61000-4-11/EN 61000-4-11

40% UT during 10/12 cycles;
70% UT during 25/30 cycles
Voltage interruptions：0% UT during
250/300 cycles

SafetyUL 61010-1:2012/R: 2018-11; CAN/CSA-C22.2 No. 61010-1:2012/A1:2018-11.
UL 61010-2-030:2018; CAN/CSA-C22.2 No. 61010-2-030:2018.

Power Supply
Input Voltage

Power

100 ～ 240 Vrms 50/60Hz
100 ～ 120 Vrms 400Hz
50 W Max (Four-channel series)
25 W Max (Two-channel series)

Mechanical (Four-channel series)
Length: 312 mm
Dimensions

Width: 132.6 mm
Height: 151 mm

WeightN.W: 2.6 kg; G.W: 3.8 kg

Mechanical (Two-channel series)
Dimensions

WWW.SIGLENT.COM
15

Length: 312 mm

SDS1000X-E Series Digital Oscilloscope
Width: 134 mm
Height:150 mm
WeightN.W:2.5 kg; G.W: 3.5 kg

Probes and Accessories
Probe

Passive

PictureModel

Specifications &Description

PP470

Bandwidth: 70 MHz, 1X/10X, 1M/10Mohm, 150V/300V

PP510

Bandwidth: 100MHz, 1X/10X, 1M/10Mohm,150V/300V

PP215

Bandwidth: 200MHz, 1X/10X, 1M/10Mohm, 150V/300V

Bandwidth: 100 KHz, Max. continuous current: 20Arms Peak current:
CP402060A
Switch Ratio: 50mV/A, 5mV/A, Accuracy: 50mV/A (0.4A -10Apk)±2%,
5mV/A(1A-60Apk) ±2%, 9V battery source
Bandwidth: 1MHz, Max. continuous current: 50Arms, Peak current:

CP4050140A
Switch Ratio: 500mV/A, 50mV/A
Accuracy: 500mV/A (20mA -14ApK)±3%±20mA , 50mV/A (200mA 100ApK)±4%±200mA, 50mV/A (100A -140ApK) ±15%max, 9V battery source
Bandwidth: 150kHz, Max. continuous current: 70Arms, Peak current:

CP4070200A
Switch Ratio: 50mV/A, 5mV/A, Accuracy: 50mV/A (0.4A -10ApK)±2% ,Current Probe

5mV/A(1A -200ApK) ±2%, 9V battery source
Bandwidth: 50 MHz, Max. continuous current: 30Arms, Peak current:
CP503050A
Switch Ratio: 100mV/A, 1V/A, Accuracy: 1V/A (±1%±1mA), 100mV/A(±1%±10mA), DC12V/1.2A power adapter
Bandwidth: 100 MHz, Max. continuous current: 30Arms, Peak current:

CP5030A

50A
Switch Ratio: 100mV/A, 1V/A, Accuracy: 1V/A (±1%±1mA), 100mV/A(±1%±10mA), DC12V/1.2A power adapter
Bandwidth: 12 MHz, Max. continuous current: 150Arms, Peak current:

CP5150300A
Switch Ratio: 100mV/A, 10mV/A, Accuracy: 100mV/A (±1%±10mA),
10mV/A (±1%±100mA), DC12V/1.2A power adapter

16
WWW.SIGLENT.COM

Bandwidth: 5 MHz, Max. continuous current: 500Arms, Peak current:
CP5500750A
Switch Ratio: 100mV/A, 10mV/A, Accuracy: 100mV/A (±1%±10mA),
10mV/A(±1%±100mA), DC12V/1.2A power adapter

DPB4080

Bandwidth: 50MHz, Differential Range: 800V (DC + Peak AC),
100X/200X/500X/1000X, Accuracy: ±1%, DC 9V/1A power adapter

DPB5150

Bandwidth: 70MHz, Differential Range: 1500V (DC + Peak AC),50X/500X
Accuracy: ±2%, DC 5V/1A USB adapter

DPB5150A
Differential Probe

Bandwidth: 100MHz, Differential Range: 1500V (DC + Peak AC),
50X/500X , Accuracy: ±2%
DC 5V/1A USB adapter

DPB5700

Bandwidth: 70MHz, Differential Range: 7000V (DC + Peak AC),
100X/1000X , Accuracy: ±2%,
DC 5V/1A USB adapter
Bandwidth: 100MHz

DPB5700A

Differential Range: 7000V (DC + Peak AC),
100X/1000X
Accuracy: ±2%
DC 5V/1A USB adapter
Bandwidth: 40MHzHigh Voltage

HPB4010Differential Range: DC 10kV, AC (rms): 7kV (sine), AC (Vpp): 20kV (Pulse)
1000X
Accuracy: ≤3%
Provides isolation between standard oscilloscope channels, isolation

Isolated front end

ISFE

betweenthe measured signal and ground. Uses USB 5V power supply, plug andplay.
The maximum input voltage allowed is up to ± 600Vpk.

STB-3 Test
Demo Board

WWW.SIGLENT.COM
17

Board

Output signals including square, sine, AM, fast edge, pulse, PWM, I2C,
CAN, LIN etc. Used in teaching and demonstrations.

SDS1000X-E Series Digital Oscilloscope

USB Isolated AWG
ModuleRack Mount

18
WWW.SIGLENT.COM

Output Sine, Square, Ramp, pulse, Noise, DC and 45 built-in
SAG1021I

waveforms. The arbitrary waveforms can be accessed and edited by
the EasyWave PC software.

SDS1X-E-RMK

The height is 4U.

Ordering Information
Ordering information
Product NameSDS1104X-E 100MHz Four Channels
SDS1204X-E 200MHz Four Channels
SDS1202X-E 200MHz Two Channels

Standard Accessories

USB Cable -1
Quick Start -1
Passive Probe -4/2
Certification -1
Power Cord -1

Optional Accessories

16 Channels Logic Analyzer (four-channel series only)

SLA1016

USB Isolated AWG Module Hardware (four channel series only)

SAG1021I

Isolated Front EndISFESTB Demo Source

STB-3

High Voltage Probe

HPB4010CP4020/CP4050/CP4070/

Current Probes

CP4070A/CP5030/CP5030A/
CP5150/CP5500

Differential Probes
Rack Mount

WWW.SIGLENT.COM
19

DPB4080/DPB5150/DPB5150A/DPB5700/DPB5700A
SDS1X-E-RMK

SDS1000X-E Series Digital Oscilloscope

20
WWW.SIGLENT.COM