SDS1000X HD
Digital Storage
Oscilloscope

Data Sheet
2022.02

EN01G

SDS1104X HD
SDS1204X HD
SDS1102X HD
SDS1202X HD

Product Overview
SIGLENT’s SDS1000X HD series highresolution digital storage oscilloscopes are
based on 2 GSa/s, 12-bit Analog-Digital
Converters and front ends with excellent noise
floor performance. They are available inbandwidths of 200 MHz and 100 MHz, havemaximum record length of 100 Mpts, and display
2/4 analog channels + 16 digital channels mixed
signal analysis ability.
The SDS1000X HD series employs Siglent’s
SPO technology with a maximum waveform
capture rate of up to 120,000 wfm/ s (normalmode, up to 500,000 wfm/s in Sequence mode),
256-level intensity grading display function plus
a color temperature display mode. It also
employs an innovative digital trigger system withhigh sensitivity and low jitter. The trigger system
supports multiple powerful triggering modes
including serial bus triggering. Tools such as
History waveform recording, Search andNavigate functions, Mask Test, Bode Plot, Power
Analysis allow for extended waveform records tobe captured, stored, and analyzed. An
impressive array of measurement and mathcapabilities, serial decoding as well as an
optional 25 MHz arbitrary waveform generator
are also features of the SDS1000X HD.
The large 10.1’’ display capacitive touch screen
supports multi-touch gestures, with the additionof user-friendly UI design, can greatly improve
the operation efficiency. It also supports mouse
control, and remote web control over LAN.

SDS1000X HD Data Sheet

Key Features
12-bit High Resolution
12-bit Analog-Digital Convertors withsample rate up to 2 GSa/s
Front ends with 70 μVrms noise floor @200 MHz bandwidth
2/4 analog channels, up to 200 MHz bandwidth
SPO technology
Waveform capture rate up to 120,000
wfm/s (normal mode), and 500,000 wfm/s(sequence mode)Supports 256-level intensity grading andcolor temperature display modesUp to 100 Mpts record length
Digital trigger system
Intelligent trigger: Edge, Slope, Pulse width,
Window, Runt, Interval, Dropout, Pattern, Video(HDTV supported) , Qualified, Nth edge, Delay,
Setup/Hold time.
Serial bus triggering and decoder, supports
protocols I2C, SPI, UART, CAN, LIN, CAN
FD(decode only), FlexRay(decode only)
Segmented acquisition (Sequence) mode,
dividing the maximum record length into multiple
segments (up to 80,000), according to trigger
conditions set by the user, with a very small
dead time between segments to capture thequalifying event
History waveform record (History) function, themaximum recorded waveform length is 80,000
framesAutomatic measurements on 50+ parameters,
supports statistics with histogram, track,
trend, Gating measurement, and measurements
on Math, History and Ref
4 Math traces (2 Mpts FFT, addition, subtraction,
multiplication, division, integration, differential,
square root, etc.), supports formula editor
Abundant data analysis functions such as
Search, Navigate, Counter, Bode Plot andPower Analysis
High Speed hardware-based Mask Test
function, with Mask Editor tool for creating userdefined masks
16 digital channels (optional)
25 MHz waveform generator(optional)
Large 10.1” TFT-LCD display with 1024 * 600
resolution; Capacitive touch screen supports
multi-touch gestures
Interfaces include: USB Hosts, USB Device(USBTMC), LAN (VXI-11/Telnet/Socket),
Pass/Fail, Trigger Out
Built-in web server supports remote control over
the LAN port using a web browser. Supports
SCPI remote control commands. Supports
external mouse and keyboard
1Models and Key Specifications
Model
Analog channels

SDS1104X HD

SDS1204X HD

SDS1102X HD

SDS1202X HD

4 + EXT(4CH Series: SDSxxx4X HD),
2 + EXT(2CH Series: SDSxxx2X HD)

Bandwidth

100 MHz

Vertical resolution

12-bitSample rate
(Max.)
Memory depth(Max.)
Waveform
capture rate
(Max.)

200 MHzOne channel mode: 2 GSa/s,
Two channel mode: 1 GSa/s,
Four channel mode: 500 MSa/s
One channel mode: 100 Mpts/ch,
Two channel mode: 50 Mpts/ch,
Four channel mode: 25 Mpts/ch
Normal mode : 120,000 wfm/s;
Sequence mode : 500,000 wfm/s

Trigger type

Edge, Slope, Pulse width, Window, Runt, Interval, Dropout, Pattern, Video,
Qualified, Nth edge, Delay, Setup/Hold time, Serial

Serial trigger anddecode(Standard)

I2C, SPI, UART, CAN, LIN, CAN FD(Decode Only), FlexRay(Decode Only)

Measurement

50+ parameters, statistics, histogram, trend, and track supported
4 tracesMath2

2 Mpts FFT, Filter, +, -, x, ÷, ∫dt, d/dt, √, Identity, Negation, Absolute, Sign, ex,
10x, ln, lg, Interpolation, MaxHold, MinHold, ERES, Average. Supports formula
editorData analysis

Search, Navigate, History, Mask Test, Counter, Bode plot, and Power Analysis

Digital channel(optional)

16-channel; maximum sample rate up to 1 GSa/s; record length up to 10 Mpts

USB AWG
module (option)

One channel, 25 MHz, sample rate of 125 MHz, wave length of 16 kpts,
isolated output

I/O

USB 2.0 Host x3, USB 2.0 Device, 10 M / 100 M LAN, External trigger,
Auxiliary output (TRIG OUT, PASS/FAIL), SBUS (Siglent MSO)

Probe (Standard)

Passivechannel

Display

10.1 TFT-LCD with capacitive touch screen (1024*600)

probe

PP510

for

each

Passivechannel

probe

PP215

for

each

SDS1000X HD Data SheetFunctions & Characteristics
12-bit High Resolution

8-bit

12-bitVertical & Horizontal Zoom along with a large memory depth make the most out of 12-bit ADC
resolution. Engineers can observe waveform overall and details simultaneously.

Low noise floor: Only 70 µVrms at 200 MHz
bandwidth

0.5% DC gain accuracy

Excellent User Interface and User Experience







SDS1000X HD Data Sheet

10.1” display with 1024x600 resolution
Capacitive touch screen, supporting multitouch gestures, can move or scale thewaveform traces quickly by finger-touch
movements, which greatly improves theoperation efficiency
Built-in WebServer supports remote control
on a web page over LAN
Supports external mouse and keyboard

3

High Waveform Update Rate

Deep Record LengthWith a waveform update rate of up to 120,000
wfm/s, the oscilloscope can easily capture
unusual or low-probability events. In Sequencemode, the waveform capture rate can reach
500,000 wfm/s.

Using hardware-based Zoom technique andrecord length of up to 100 Mpts, users can selecta slower timebase without compromising thesample rate, and then quickly zoom in to focus on
the area of interest.

Multiple Trigger Functions

Edge, Slope, Pulse width, Video, Window, Runt,
Interval, Dropout, Pattern, Qualified, Nth edge,
Delay, Setup/Hold time, and serial trigger.

4

SDS1000X HD Data Sheet

Advanced Math Function

In addition to the traditional (+, -, X, /) operations,
FFT, integration, differential, square root, andmore are supported. Formula Editor is available
for more complex operations. 4 math traces areavailable.

Hardware-accelerated FFT supports up to 2 Mpts
operation. This provides high-frequency resolution
with a fast refresh rate. The FFT function also
supports a variety of window functions so that it
can adapt to different spectrum measurement
needs. Three modes (Normal, Average, and Maxhold) can satisfy different requirements forobserving the power spectrum. Auto peakdetection and markers are supported.

Measurements of a Variety of Parameters

Parameter Statistics Function

Parameter measurements include 4 categories:
horizontal, vertical, miscellaneous, and CH delay
providing a total of 50+ different types ofmeasurements. Measurements can be performed
within a specified gate period. Measurements on
Math, Reference, and History frames aresupported.

Statistics show the current value, maximum value,
minimum value, standard deviation, and meanvalue of up to 12 parameters simultaneously. A
histogram is available to show the probability
distribution of a parameter. Trend and Track areavailable to show the parameter value vs. time.
For horizontal parameters such as period, allresults are extracted from a frame, instead of justcalculating the first one. This accelerates statistics
on horizontal measurements much more andenables distribution observation in a frame using
Histogram and Track.

SDS1000X HD Data Sheet

5

History Mode

Sequence Mode

History function can record up to 80,000 frames ofwaveforms.
The
recordingisexecuted
automatically so that the customer can playback
the history waveforms at any time to observe
unusual events and quickly locate the area ofinterest using the cursors or measurements.

Segmented memory collection will store thewaveform into multiple memory segments (up to
80,000) and each segment will store a triggered
waveform as well the dead time information. Theinterval between segments can be as small as 2
μs. All of the segments can be played back using
the History function.

Search and Navigate

Serial Bus Decode

The oscilloscope can search events specified by
the user in a frame. Events flagged by the Search
can be recalled automatically using Navigate. It
can also navigate by time (delay position) andhistory frames.

Display the decoded characters through theevents list. Bus protocol information can be quickly
and intuitively displayed in tabular form. I2C, SPI,UART, CAN, LIN, CAN FD, FlexRay aresupported.

Hardware-based High-Speed Mask Test Function

6

SDS1000X HD Data Sheet

The oscilloscope utilizes a hardware-based Mask
Test function, performing up to 80,000 Pass / Faildecisions each second. It is easy to generateuser-defined test templates to provide trace mask
comparisons, making it suitable for long-term
signal monitoring or automated production line
testing.

Built-in Mask Editor application helps to createcustom masks.

Bode PlotPower Analysis (Optional)

SDS1000X HD can control the USB AWG module
or control an independent SIGLENT SDG
instrument, scan a devices amplitude and phasefrequency response, and display the data as a
Bode Plot. There is also a Vari-level Mode for
accurately measuring Power Supply Control Loop
Response (PSRR).

The Power Analysis option provides a full suite of
power measurements and analysis, which greatly
improve the measurement efficiency in switching
power supplies and power devices design.

Digital Channels / MSO (Optional)

USB AWG module (Optional)

Four analog channels plus 16 digital channels
enable users to acquire and trigger the waveforms
then analyze the pattern, simultaneously with one
instrument.

The USB waveform generator can output
waveforms with up to 25 MHz frequency and ±3 V
amplitude. Six basic waveforms together withmultiple types of predefined waveforms and asuser-defined arbitrary waveforms are supported.

SDS1000X HD Data Sheet

7

Specifications
All specifications are not guaranteed unless the following conditions are met:
• The oscilloscope calibration period is current• The oscilloscope has been working continuously for at least 30 minutes at the specified temperature (18℃ ~28℃)
Acquire (analog)

Waveform updaterate

One channel mode: 2 GSa/s,
Two channel mode: 1 GSa/s,
Four channel mode: 500 MSa/s
One channel mode: 100 Mpts/ch,
Two channel mode: 50 Mpts/ch,
Four channel mode: 25 Mpts/ch
Normal mode : up to 120,000 wfm/s
Sequence mode : up to 500,000 wfm/s

Intensity grading

256-level

Peak detect

2 ns

Sequence

Up to 80,000 segments, interval between triggers = 2 μs min.

History

Up to 80,000 frames

Interpolation

sinx/x, x

Sample rate

Memory depth

Vertical (analog)

Rise time@50 Ω(typical)

4 + EXT (4CH Series: SDSxxx4X HD),
2 + EXT (2CH Series: SDSxxx2X HD)
SDS1104X HD, SDS1102X HD: 100 MHz
SDS1204X HD, SDS1202X HD: 200 MHz
DC - 10% (BW): ±1 dB
10% - 50% (BW): ±2 dB
50% - 100% (BW): +2 dB / -3 dB
Typical 3.5 ns (SDS1104X HD, SDS1102X HD)
Typical 1.8 ns (SDS1204X HD, SDS1202X HD)

Resolution

12-bitENOB *1 (typical)

8.4-bit

Noise floor*2(rms, @50 Ω,
typical,1 mV/div)

70 μV(Full Bandwith)Range

8 divisions

Vertical scale(probe 1X)

1 MΩ: 0.5 mV/div – 10 V/div
50 Ω: 0.5 mV/div – 1 V/div

ChannelBandwidth(-3 dB) @ 50 Ω
Flatness@50 Ω

8

SDS1000X HD Data Sheet0.5 mV/div ~ 4.95 mV/div:
DC gain accuracy±1.5 %±0.5 % (typical)

5 mV/div ~ 10 V/div:±1 % (max)Offset accuracy± (0.5% of the offset setting + 0.5% of full scale + 1 mV)

Offset range(probe 1X)

1 MΩ:
0.5 mV/div ~ 5 mV/div: ±1.6 V;
5.1 mV/div ~ 10 mV/div: ±4 V;
10.2 mV/div ~ 100 mV/div: ±8 V;
102 mV/div ~ 1 V/div: ±80 V;
1.02 V/div ~ 10 V/div: ±400 V

Bandwidth limit

Hardware Bandwidth limit: 20 MHz, 200 MHz

Low frequency
response (ACcoupling -3 dB)
Overshoot (150ps edge @50 Ω,
typical)Coupling

2 Hz (typical)

10%
DC, AC, GND

Max. input
voltage

1 MΩ: (1 MΩ±2%) || (17 pF±2 pF)
50 Ω: 50 Ω±1%
1 MΩ ≤ 400 Vpk (DC + AC), DC ~ 10 kHz
50 Ω ≤ 5V rms

SFDR≥ 35dBc

CH to CH
Isolation (@50Ω)DC ~ Max BW: > 40dB

Probe attenuation

1X, 10X, 100X, Custom

Impedance

50 Ω:
0.5 mV/div ~ 5 mV/div: ±1.6 V;
5.1 mV/div ~ 10 mV/div: ±4 V;
10.2 mV/div ~ 100 mV /div: ±8 V;
102 mV/div ~ 1 V/div: ±10 V;* 1:24.99 MHz input , -0.25 dBFS , 20 mV/div , 50 Ω input impedance* 2:Use the “Stdev” measurement
Horizontal
Time scale

1 ns/div – 1000 s/div

Range

10 divisions

Display mode

Y-T, X-Y, Roll

Roll mode≥ 50 ms/div

Skew (CH1~CH4)< 100 psTime baseaccuracy±25 ppm

SDS1000X HD Data Sheet

9Trigger
Mode

Auto, Normal, Single

Level

Internal : ± 4.1 div from the center of the screen
EXT : ± 0.61 V
EXT/5 : ± 3.05 V

Hold off range

By time : 8 ns ~ 30 s (8 ns step)

Coupling

AC
DC
LFRJ
HFRJ
Noise RJ

Coupling
frequency
response(CH1 ~ CH4)

DC : Passes all components of the signal
AC : Blocks DC components and attenuates signals below 8 Hz
LFRJ : Blocks the DC component and attenuates the low-frequency
components below 2 MHz
HFRJ : Attenuates the high-frequency components above 2.4 MHz

Coupling
frequency
response(EXT)

DC : Passes all components of the signal
AC : Blocks DC components and attenuates signals below 18 Hz
LFRJ : Attenuates the frequency components below 7.5 kHz
HFRJ : Attenuates the frequency components above 250 kHz

Accuracy (typical)

CH1 ~ CH4: ±0.2 div
EXT : ±0.3 div

Sensitivity

CH1 ~ CH4 :DC ~ Max BW : 0.6 div

EXT :200 mVpp, DC ~ 200 MHz

EXT/5 :

1 Vpp, DC ~ 200 MHz

Jitter

CH1 ~ CH4 : < 100ps
EXT : < 200 ps rms

Displacement

Pre-Trigger : 0 ~ 100% memory
Delay-Trigger : 0 ~ 10,000 div

Edge Trigger
SourceCH1 ~ CH4 / EXT / (EXT/5) / AC Line / D0 ~ D15

Slope

Rising, Falling, Alternating

Slope Trigger
SourceCH1 ~ CH4

Slope

Rising, Falling

Limit range<, >, in range, out of range

Time range

2 ns ~ 20 s, Resolution = 1 ns

Pulse Width Trigger
SourceCH1 ~ CH4 / D0 ~ D15

Polarity+width, -width

Limit range<, >, in range, out of range

Time range

2 ns ~ 20 s, Resolution = 1 ns

10

SDS1000X HD Data SheetVideo Trigger
SourceCH1 ~ CH4

Standard

NTSC, PAL, 720p/50, 720p/60, 1080p/50, 1080p/60, 1080i/50, 1080i/60, Custom

Synchronization

Any, SelectTrigger condition

Line, Field

Window Trigger
SourceCH1 ~ CH4

Window type

Absolute, Relative

Interval Trigger
SourceCH1 ~ CH4 / D0 ~ D15

Slope

Rising, Falling

Limit range<, >, in range, out of range

Time range

2 ns ~ 20 s, Resolution = 1 ns

Dropout Trigger
SourceCH1 ~ CH4 / D0 ~ D15

Timeout type

Edge, State

Slope

Rising, Falling

Time range

2 ns ~ 20 s, Resolution = 1 ns

Runt Trigger
SourceCH1 ~ CH4

Polarity

Positive, Negative

Limit range<, >, in range, out of range

Time range

2 ns ~ 20 s, Resolution = 1 ns

Pattern Trigger
SourceCH1 ~ CH4 / D0 ~ D15

Pattern setting

Don’t Care, Low, High

Logic

AND, OR, NAND, NOR

Limit range<, >, in range, out of range

Time range

2 ns ~ 20 s, Resolution = 1 ns

Qualified Trigger
Type
Qualified source
Edge source
Qualify state
Qualify edge
Limit range
Time range
Nth Edge Trigger
SourceSlope
Idle time

SDS1000X HD Data Sheet

State, State with Delay, Edge, Edge with Delay
CH1 ~ CH4
CH1 ~ CH4
Low, High
Rising, Falling<, >, in range, out of range
2 ns ~ 20 s, Resolution = 1 ns
CH1 ~ CH4
Rising, Falling
8 ns ~ 20 s, Resolution = 1 ns

11

Edge number
Delay Trigger
Source A
Source B
Slope
Limit range
Time range
Setup/Hold Trigger
Option
CLK source
CLK slope
Data source
Data state
Limit range
Time range

1 ~ 65535
CH1 ~ CH4
CH1 ~ CH4
Rising, Falling<, >, in range, out of range
2 ns ~ 20 s, Resolution = 1 ns
Setup, Hold
CH1 ~ CH4
Rising, Falling
CH1 ~ CH4
Low, High<, >, in range, out of range
2 ns ~ 20 s, Resolution = 1 ns

Serial Trigger
SourceCH1 ~ CH4 / D0 ~ D15

ProtocolI2C, SPI, UART, CAN, LIN

I2CType : Start, Stop, Restart, No Ack, EEPROM, Address & Data, Data Length

SPI

Type : Data

UART

Type : Start, Stop, Data, Parity Error

CAN

Type : All, Remote, ID, ID+Data, Error

LIN

Type : Break, Frame ID, ID+Data, Error

12

SDS1000X HD Data Sheet

Serial Decoder
Decoders

2

Threshold-4.5 ~ 4.5 div

List

1 ~ 7 lines

Decoder type

Full duplex

2

IC
SourceCH1 ~ CH4 / D0 ~ D15

SignalSCL, SDA

Address

7-bit, 10-bit

SPI
SourceCH1 ~ CH4 / D0 ~ D15

Signal

CLK, MISO, MOSI, CS

Edge selectRising, Falling

Chip selectActive high, Active low, Clock timeout

Bit Order

LSB, MSB

UART
SourceCH1 ~ CH4 / D0 ~ D15

SignalRX, TX

Data width

5-bit, 6-bit, 7-bit, 8-bit

Parity check

None, Odd, Even, Mark, Space

Stop bit

1-bit, 1.5-bit, 2-bit

Idle level

Low, HighBit order

LSB, MSB

CAN
SourceCH1 ~ CH4 / D0 ~ D15

LIN
LIN version

Ver 1.3, Ver 2.0

SourceCH1 ~ CH4 / D0 ~ D15

Baud rate

600 bps, 1200 bps, 2400 bps, 4800 bps, 9600 bps, 19200 bps, Custom

CAN FD
SourceNominal baud rate
Data baud rate
FlexRay
SourceBaud rate

CH1 ~ CH4
10 kbps, 25 kbps, 50 kbps, 100 kbps, 250 kbps, 1 Mbps, Custom
500 kbps, 1 Mbps, 2 Mbps, 5 Mbps, 8 Mbps, 10 Mbps, Custom
CH1 ~ CH4
2.5 Mbps, 5 Mbps, 10 Mbps, Custom

Measurement
Automatic Measurement
SourceSDS1000X HD Data Sheet

CH1 ~ CH4, D0 ~ D15, Z1 ~ Z4, F1 ~ F4, Ref, History

13

Mode

Simple, Advanced

Range

ScreenGated: inside screen, definable with separate Gate cursors

Custom threshold

Upper, Middle, Lower

No. ofmeasurements

Display 12 measurements at the same time (Display mode = M2)

Verticalparameters

Max, Min, Pk-Pk, Top, Base, Amplitude, Mean, Cycle Mean, Stdev, Cycle Stdev,
RMS, Cycle RMS, Median, Cycle Median, FOV, FPRE, ROV, RPRE,
Level@Trigger

Horizontal
parameters

Period, Frequency, Time@max, Time@min, +Width, -Width, 10-90%Rise time,
90-10%Fall time, Rise time, Fall time, +Burst Width, -Burst Width, +Duty Cycle,-Duty Cycle, Delay, Time@Middle, Cycle-Cycle jitter

Miscellaneous
parameters+Area@DC, -Area@DC, Area@DC, Absolute Area@DC, +Area@AC, Area@AC, Area@AC, Absolute Area@AC, Cycles, Rising Edges, Falling Edges,
Edges, Positive pulses, Negative pulses, Positive Slope, Negative Slope

Delay parameters

Phase, FRFR, FRFF, FFFR, FFFF, FRLR, FRLF, FFLR, FFLF, Skew, Tsu@R,
Tsu@F, Th@R, Th@F

Statistics

Current, Mean, Min, Max, Pk-Pk, Sdev, Count, Histogram, Trend, Track

Statistics countUnlimited, 1~1024

Cursors
SourceCH1 ~ CH4, D0 ~ D15, F1 ~ F4, Ref

TypeManual : Time X1, X2, (X1-X2), (1/ΔT); Vertical Y1, Y2, (Y1-Y2)
Track: Time X1, X2, (X1-X2)
Measure: indicates the measurement on specific parameter

Math
Trace

F1 ~ F4

SourceCH1 ~ CH4, Z1 ~ Z4, F1 ~ F4

Operation

FFT

FFT, Filter, +, -, x, ÷, ∫dt, d/dt, √, Identity, Negation, |x|, Sign, ex, 10x, ln, lg,
Interpolation, Max hold, Min hold, ERES, Average, Formula Editor
Length: 2 Mpts, 1 Mpts, 512 kpts, 256 kpts, 128 kpts, 64 kpts, 32 kpts, 16 kpts,
8 kpts, 4 kpts, 2 kpts, 1 kpts
Window: Rectangle, Blackman, Hanning, Hamming, Flattop
Display: Full Screen, Split, Exclusive
Mode: Normal, Max hold, Average
Tools: Peaks, Markers

Analysis
SearchSourceCH1 ~ CH4, History

Mode

Edge, Slope, Pulse, Interval, Runt

Copy setting

Copy from trigger, Copy to trigger

14

SDS1000X HD Data Sheet

Navigate
Type

Search event, Time, History frame

Mask Test
SourceCH1 ~ CH4 , Z1 ~ Z4

Mask creating

Auto (Create mask), Customized (Mask Editor)

Mask test speed

Up to 80,000 frames/s

Bode Plot
SourceCH1 ~ CH4

Supported signal
sources

SAG1021I (Connection: USB),
SDG series waveform generators (Connection: USB, LAN)

Sweep type

Simple, Vari-level

Frequency

Mode: Linear, Logarithmic
Range: 10 Hz ~ 120 MHz

MeasureUpper cutoff frequency, Lower cutoff frequency, Bandwidth, Gain margin,
Phase margin

Power Analysis (optional)
MeasurePower quality, Current Harmonics, Inrush current, Switching loss, Slew rate,
Modulation, Output ripple, Turn on/turn off, Transient response, PSRR, SOA,
Efficiency(four channel series only)

Counter
SourceCH1 ~ CH4

Frequency
resolution

7 digits

Totalizer

Counter on edges, supports Gate and Trigger

Digital Channels (optional)Max. sampling
rate
Memory depth
Min. detectablepulse width
Level group
Level range
Logic type
Skew

1 GSa/s
10 Mpts/ch
3.3 ns
D0 ~ D7, D8 ~ D15-8 V ~ +8 V
TTL, CMOS, LVCMOS3.3, LVCMOS2.5, Custom
D0 ~ D15: ±1 sampling interval
Analog to Digital(Triggered On Digital): ± (1 sampling interval +1 ns)
Digital to Analog(Triggered On Analog): ±4 ns

USB AWG Module (optional)
Channels1Max. output
frequency

25 MHz

SDS1000X HD Data Sheet

15

Sampling rate
Frequency
resolution
Frequency
accuracy

125 MSa/s
1 μHz±50 ppm

Vertical resolution

14 bitAmplitude range-1.5 V ~ +1.5 V (into 50 Ω)-3 V ~ +3 V (into High-Z)

Waveforms

Sine, Square, Ramp, Pulse, DC, Noise, 45 Arbitrary

Outputimpedance

50 Ω ± 2%

Protection

Over voltage protection, Current limit

Sine
Frequency
Offset accuracy(10 kHz)
Amplitude
flatness
SFDR
Harmonic
distortion

1 μHz ~ 25 MHz±(1%*offset setting value +3 mVpp)±0.3 dB, compare to 10 kHz, 2.5 Vpp into 50 Ω
DC ~ 1 MHz
1 MHz ~ 5 MHz
5 MHz ~ 25 MHz
DC ~ 5 MHz
5 MHz ~ 25 MHz-60 dBc-55 dBc-50 dBc-50 dBc-45 dBc

Square/Pulse
Frequency

1 μHz ~ 10 MHz

Duty cycle

1% ~ 99%

Edge< 24 ns (10% ~ 90%)

Overshoot< 3% (typical , 1 kHz, 1 Vpp)

Pulse width> 50 ns

Jitter (cycle-cycle)< 500 ps + 10 ppm

Ramp
Frequency

1 μHz ~ 300 kHz

Linearity< 0.1% of Pk-Pk (typical, 1 kHz, 1 Vpp, 50% symmetry)

Channels

0% ~ 100%

DC
Offset range±1.5 V (into 50 Ω)±3 V (into Hi-Z)

Accuracy±(|setting value|*1% + 3 mV)

NoiseBandwidth (-3 dB)> 25 MHz

Arb
Frequency

1 μHz ~ 5 MHz

Wave memory

16 kpts

16

SDS1000X HD Data SheetSample rate

125 MSa/s

Wave import

From EasyWaveX, from U-disk, directly from waveform data of analog channels

I/O
Front

USB 2.0 Host x2,
SBUS: Siglent MSO,
Calibration Signal: 1 kHz, 3 V Square

Rear

USB 2.0 Host,
USB 2.0 Device,
LAN: 10/100MbaseT (RJ45),
External Trigger , EXT: ≤ 1.5 Vrms , EXT/5: ≤ 7.5Vrms,
Auxiliary Output: TRIG OUT (3.3 V LVCMOS), PASS/FAIL OUT (3.3 V TTL)

Display
Display type10.1 TFT LCD with capacitive touch screen

Resolution

1024×600

Contrast (typical)

500:1

Backlight (typical)

500 nit

Display SettingRange

8 x 10 grid

Display typeDot, Vector

Persistence timeOFF, 1 s, 5 s, 10 s, 30 s, infinite

Color display

Normal, Color

Language

Simplified Chinese, Traditional Chinese, English, French, Japanese, German,
Spanish, Russian, Italian, Portuguese

Built-in help
system

Simplified Chinese, English

Environmental
Temperature

Operating: 0 °C ~ 50 °C
Non-operating: -30 °C ~ 70 °C

Humidity

Operating: 5% ~ 90%RH, 30 °C, degraded to 50%RH at 50 °C
Non-operating: 5% ~ 95%

AltitudeOperating: ≤ 3,000 m, 25 °C
Non-operating: ≤ 15,000 m

Electromagnetic
compatibility

SDS1000X HD Data Sheet

Meets EMC directive (2014/30/EU), meets or exceeds IEC 613261:2012/EN61326-1:2013 (Basic)
Conducted disturbance

CISPR 11/EN
55011

CLASS A group 1
150 kHz-30 MHz

17

Radiated disturbance

CISPR 11/EN
55011

CLASS A group 1
30 MHz-1 GHz

Electrostatic discharge(ESD)

IEC 61000-4-2/EN
61000-4-2

4.0 kV (Contact),8.0 kV (Air)IEC 61000-4-3/EN
61000-4-3

10 V/m (80 MHz to 1 GHz);
3 V/m (1.4 GHz to 2 GHz);
1 V/m (2.0 GHz to 2.7 GHz)Radio-frequency
electromagnetic field
Immunity
Electrical fast transients(EFT)
SurgesRadio-frequency
continuous conducted
Immunity

SafetyRoHS

IEC 61000-4-4/EN
61000-4-4
IEC 61000-4-5/EN
61000-4-5
IEC 61000-4-6/EN
61000-4-6

2kV (Input AC Power Ports)
1kV (Line to line)
2kV (Line to ground)
3 V, 0.15-80 MHz

Voltage Dips:
0% UT during 1 cycle;
40% UT during 10/12 cycles;
Voltage dips andIEC 61000-4-11/
70% UT during 25/30 cycles
interruptions
EN 61000-4-11
Voltage interruptions:
0% UT during 250/300
cyclesUL 61010-1:2012/R: 2018-11; CAN/CSA-C22.2 No. 61010-1:2012/A1:2018-11.
UL 61010-2-030:2018; CAN/CSA-C22.2 No. 61010-2-030:2018.
EU 2015/863

Power Supply
Input voltage &frequency

100 ~ 240 Vrms 50/60 HzPowerconsumption

120 W max., 70 W typical, 4 W typical in standby mode

Mechanical
Dimensions

Length × Height × Width = 317.2 mm × 236.0 mm × 149.0 mm(including knobs and supporting legs)

Weight

Net Weight 4.1 kg, Gross Weight 5.6 kg

18

SDS1000X HD Data SheetOrdering Information
Model

Description

SDS1204X HD

200 MHz, 2 GSa/s, 4CH

SDS1104X HD

100 MHz, 2 GSa/s, 4CH

SDS1202X HD

200 MHz, 2 GSa/s, 2CH

SDS1102X HD

100 MHz, 2 GSa/s, 2CH

Standard Accessories

Quantity

USB cable

1

Quick start

1

Passive probe

1/channel

Certificate of calibration

1

Power cord

1

Optional Accessories

Part No.

USB Isolated AWG Module Hardware

SAG1021I

16 Channels Logic Analyzer

SLA1016

Power Analysis Software

SDS1000XHD-PA

Power Analysis Deskew Fixture

DF2001A

Accessories
Accessories

PictureModel

Specifications &Description

STB-3 Test
Board

Output signals including square, sine,AM, fast edge, pulse, PWM, I2C, CAN,
LIN etc. Used in teaching anddemonstrations.

USB Isolated
AWG Module

SAG1021I

Output Sine, Square, Ramp, pulse,
Noise, DC and 45 built-in waveforms. The
arbitrary waveforms can be accessedand edited by the EasyWave PC
software.

16 Channels
Logic
Analyzer

SLA1016

Provides 16 digital channels byconnecting the SBUS interface.

Demo Board

SDS1000X HD Data Sheet

19PowerAnalysis
Deskew
Fixture20

DF2001A

Calibrates the phase between the current
and voltage probes.

SDS1000X HD Data SheetAbout SIGLENTSIGLENT is an international high-tech company, concentrating on R&D, sales,
production and services of electronic test & measurement instruments.
SIGLENT first began developing digital oscilloscopes independently in 2002.
After more than a decade of continuous development, SIGLENT has extended
its product line to include digital oscilloscopes, isolated handheld
oscilloscopes,

function/arbitrary

waveform

generators,

RF/MWsignalgenerators, spectrum analyzers, vector network analyzers, digital multimeters,
DC power supplies, electronic loads and other general purpose test
instrumentation. Since its first oscilloscope was launched in 2005, SIGLENT
has become the fastest growing manufacturer of digital oscilloscopes. We
firmly believe that today SIGLENT is the best value in electronic test &measurement.

Headquarters:
SIGLENT Technologies Co., Ltd
Add: Bldg No.4 & No.5, Antongda Industrial
Zone, 3rd Liuxian Road, Bao'an District,
Shenzhen, 518101, China
Tel: + 86 755 3688 7876
Fax: + 86 755 3359 1582
Email: sales@siglent.com
Website: int.siglent.com

North America:
SIGLENT Technologies America, Inc
6557 Cochran Rd Solon, Ohio 44139

Tel: 440-398-5800
Toll Free: 877-515-5551
Fax: 440-399-1211
Email: info@siglentna.com
Website: www.siglentna.com

Europe:Follow us on
Facebook: SiglentTech

SIGLENT Technologies Germany GmbH
Add: Staetzlinger Str. 70
86165 Augsburg, Germany
Tel: +49(0)-821-666 0 111 0
Fax: +49(0)-821-666 0 111 22

Email: info-eu@siglent.com
Website: www.siglenteu.com

SDS1000X HD Data Sheet21