DIGITAL MULTIMETER

PC7000APPLICATIONS AND FEATURES

This instrument is a portable multimeter designed for themeasurement of low-voltage circuit.
This is used at small communications equipments, homeelectric appliances, voltage of lump line, various types ofbatteries, and circuit analyses with additional functions.

●AC True RMS / 50000 count(500000 count for DCV)
●0.03% best accuracy●Dual display capability with Backlight to simultaneously display
voltage/current/frequency and AC/DC components of voltage/current●High speed bar graph●Low Pass Filter(LPF) useful for measurement of Variable Frequency Drive(VFD)●Current(mA/μA)% 4-20mA measurement●Conductance measurement / Capacitance measurement●Temperature measurement(with K type thermocouple sensor : -50℃〜1000℃)●Frequency measurement(AC sine wave only)●Logic frequency measurement●Duty cycle measurement●MAX, MIN, AVE value hold / Data hold / Range hold
●Relative value●Auto power save(17min after the last operation)(cancelable)●Backlight●Protective holster with wall hanger and lead holder / Tilt stand●PC Link system with optical Link USB cable (KB-USB7) and software(PC Link7)*KB-USB7 and PC Link7 are optional accessories

SPECIFICATIONSDCVACV
DCA
ACA
Resistance
Capacitance
Temperature
Frequency
Logic frequency
Duty cycledBmContinuity
Diode test

Best accuracy
Resolution
Measuring range±(0.03%+2)
0.01mV
500m/5/50/500/1000V±(0.5%+40)
0.01mV
500m/5/50/500/1000V±(0.1%+20)
0.01μA
500μ/5000μ/50m/500m/5/10A±(0.6%+40)
0.01μA
500μ/5000μ/50m/500m/5/10A±(0.2%+6)
0.01Ω
500/5k/50k/500k/5M/50MΩ/99.99nS *1±(0.8%+3)*2
0.01nF50n/500n/5μ/50μ/500μ/5m/25mF
0.1℃-50℃〜1000℃(thermocouple K type)±(0.3%+2)0.001Hz
10Hz〜200kHz±(0.02%+4)0.001Hz
5Hz〜2MHz±(0.002%+4)
0.01%
0.1%〜99.99%±(3d/kHz+2)
0.01dB-29.83dBm〜54.25dBm±(0.25dB+2)
Buzzer sounds at between 20Ω and 200Ω Open voltage: approx. 1.3V
Open voltage: approx. 3V

Input impedance
10MΩ

*1 nS(Conductance): High-value resistance of Giga-Ohms for leakage measurements. Conductance is the inverse of Resistance, that is S=1/Ω or nS=1/GΩ
*2 Accuracy of film capacitor or equivalent with low leakage.

A battery for monitoring has been installed prior to shipment from the factory. It may be discharged before theexpiration of the described battery life. This battery is used to check the functions and performance of theproduct.Specifications and external appearance of the product described above may be revised for modification
without prior notice.

Distributed byDempa Bldg, 4-4 Sotokanda 2-Chome, Chiyoda-Ku, Tokyo 101-0021 Japan
Tel:+81-3-3251-0941 Fax:+81-3-3256-9740

www.sanwa-meter.co.jp

Operation method
AC measuring method
DisplayRange selection
Overload
PolarityΔ-Σ
True RMS
50000 count and 500000 countAuto and manual ranges"O.L" is displayed
Automatic selection(only " - " is displayed whennegative voltage is inputted.)
Battery low warning
Battery() mark lights when the internal
battery's power is below approx. 7V
Numeric part : 5 times/sec(50,000 count mode),
Sampling rate
1.25 times/sec(500,000 count mode)
Bar graph part : 60 times/sec
Approx. 48mW Approx. 0.45mW(Auto power save)
Power consumption
V : 40Hz〜3kHz, 3kHz〜20kHz(below 99.99V)
BandwidthA : 40Hz〜1kHz
1000V
EN61010-1 CAT.II
600V
EN61010-1 CAT.III
11A/1000V IR20kA Φ10x38mm
6LR61(9V) x 1
Fuse / Battery
0.4A/1000V IR30kA Φ6.3x32mm
Size / Mass
H184 x W86 x D52mm / 430g
Standard accessories included Test Lead(TL-23a), Holster(H-700),
Thermocouple K type (K-250PC), Instruction Manual