SHA850A
Spectrum & Network
Analyzer

DataSheet
EN_02B

SHA800A Analyzer DataSheetGeneral Description
The SIGLENT SHA850A, a handheld portable spectrum analyzer and cable-and-antenna analyzer, is a
powerful and flexible tool for those field and outdoor RF applications. Including communicationengineering, telecom operation and maintenance, radio management, factory production, education andteaching and many other fields.

With a frequency range up to 7.5 GHz, the analyzer delivers reliable automatic measurements andmultiple modes of operation. A spectrum analyzer, including built-in amplifier and independent signal
source, fast scanning speed, high sensitivity, can achieve broadcast monitoring, channel power scanning,
wireless interference location, power monitoring, electromagnetic compatibility, and other functions. A
cable and antenna tester including built-in DC voltage bias, with a 1-path-2-port vector network analysis
function, can measure TDR, VSWR, port matching debugging, insertion loss measurement, tower
amplifier debugging, cable fault location, Smith chart, etc.

Features and Benefits
Spectrum Analyzer Frequency Range from 9 kHz up to 7.5 GHz, -165 dBm/Hz Displayed Average
Noise Level (Typ.), -104 dBc/Hz.@10 kHz Offset Phase Noise (1 GHz, Typ.), 1 Hz up to 3 MHz
Resolution Bandwidth (RBW), Preamplifier and independent signal source up to 7.5 GHz, GPS
positioning and logging
Cable and Antenna Test Frequency Range from 100 kHz up to 7.5 GHz, Distance To Fault and Time
Domain Analysis
Optional Vector Network Analyzer, Bias out up to 32VDC
Optional Analog and Digital Modulation Analysis, IQ Data Acquisition, Indoor and outdoor map
Typical working time 4 hours, 3.2 kg net weight, 8.4 lnch Multi-Touch Screen , Mouse and Keyboard
supported
Model

SHA851A

SHA852A

Spectrum Analyzer
Cable and Antenna Test

9 kHz~3.6 GHz
100 kHz~3.6 GHz

9 kHz~7.5 GHz
100 kHz~7.5 GHz

DataSheet1

Design Features

8.4 inch multi-touch screen and full keyboard control

GPS Location and trace log recorder, sync 10MHz reference clock

Interference analysis with directional antenna

2

DataSheetOutdoor and Indoor map, GPS logging

Channel Power and ACPR measurement

DataSheet3

Cable and Antenna Test based on Timing Domain Analysis

Vector Network Analyzer S11 and S21 measurement, Multi Formats Overlay Display

4

DataSheetAM/FM/PM analog modulation analysis, and ASK/FSK/PSK/MSK/QAM digitalmodulation analysis

DataSheet5

Accessories

6

Utility Kit

Near Field Probe Set

50Ω Calibration Kit

GPS Antenna

Portable Bag

Directional Antenna Kit

DataSheetSpecifications
Specifications are valid under the following conditions: The instrument is within the calibration period,has been stored between 0 and 50°C for at least 2 hours prior to use, and has been powered on andwarmed up for at least 60 minutes. The specifications include the measurement uncertainty, unlessotherwise noted.

Specifications: All products are guaranteed to meet published specifications when operating at room
temperature (approximately 25°C), unless otherwise noted.

Typical: Performance deemed typical implies that 80 percent of the measurement results will meet the
typical published performance with a 95th percentile confidence level at room temperature(approximately 25°C). Typical performance is not warranted and does not include measurement
uncertainty.

Nominal: The expected performance or design attribute.

DataSheet7

Spectrum Analyzer
Frequency and Time Characteristic
Frequency
SHA851A

SHA852A

Frequency range

9 kHz~3.6 GHz

9 kHz~7.5 GHz

Frequency resolution

1 Hz

Frequency Span
Range

0 Hz, 100 Hz to Max Frequency

Accuracy± Span / (number of display points - 1)

Internal Reference Source
Reference frequency
Reference frequency accuracy /uncertainty
Initial calibration accuracy

10.000000 MHz± [(time since last adjustment × frequency aging rate) + temperature stability +initial calibration accuracy]<1 ppmTemperature stability<1 ppm/year, 0 °C ~50 °C

Frequency aging rate

<0.5 ppm/first year, 3.0 ppm/20 years

Accuracy, synced to GPS±0.01 ppm

Accuracy, unsynced to GPS±0.4 ppm

MarkerMarker resolution
Marker uncertainty
Frequency Counter resolution

Span / (number of display points - 1)
± [frequency indication × reference frequency uncertainty + 1% × span + ½ *marker resolution + 1 Hz]
0.1 HzBandwidths
Resolution bandwidth (-3dB)

1 Hz ~ 3 MHz, in 1-3-10 sequence

Resolution filter shape factor< 4.8 : 1 (60 dB:3 dB), Gaussian-like

RBW uncertainty<5%

Video bandwidth (-3dB)

1 Hz ~ 10 MHz, in 1-3-10 sequence

VBW uncertainty<5%

Sweep and Trigger
Sweep time
RBW

1 ms to 4000 s
Sweep

3 kHz ~ 3 MHz

FFT

1 Hz ~ 10 kHz

Sweep points

201~10001

Sweep rule

Single, Continuous

Trigger source

Free, Video, External, Periodic

External trigger

5V TTL level, Rising edge/Falling edge

Amplitude Accuracy and Range Specifications

8

DataSheetAmplitude and Level

Reference level

DANL to +10 dBm, 100 kHz ~ 1 MHz, Preamp off
DANL to +20 dBm, 1 MHz ~ 7.5 GHz, Preamp off-200 dBm to +30 dBm, 1 dB steps

Preamplifier

25 dB (nom.)

Input attenuation

0~50 dB, 2 dB steps

Maximum input DC voltage+/- 50 VDC

Maximum damage level

33 dBm, fc ≥ 10 MHz, att > 20 dBm, preamp off, in 3 minutes

Measurement range

Level Display
Logarithmic level axis

1 dB to 200 dB

Linear level axis

0% to 100% (reference level)

Units of level axis

dBm, dBmV, dBμV, dBμA, Volt, Watt

Number of traces

6

Trace detectors

Positive-peak, Negative-peak, Sample, Normal, Average(Voltage/RMS/Video)

Trace functions

Clear write, Max Hold, Min Hold, View, Blank, Average, MathField Strength Measurement
Antenna factor
Unit

DataSheet4
dBm/m2, dBw/m2, dBV/m, dBmV/m, dBpV/m, V/m, W/m2, W/cm2, A/m

9

Displayed Average Noise Level (DANL)SHA851A

SHA852A

20 °C to 30 °C, att = 0 dB, RBW = 1 Hz, sample detector, trace average > 50, Source off

Preamp off

Preamp on

100 kHz ~1 MHz-125 dBm, -136 dBm (typ.)-125 dBm, -136 dBm (typ.)1 MHz~600 MHz-140 dBm, -147 dBm (typ.)-140 dBm, -147 dBm (typ.)600 MHz~1.8 GHz-139 dBm, -145 dBm (typ.)-139 dBm, -145 dBm (typ.)1.8 GHz~3.2 GHz-134 dBm, -140 dBm(typ.)-134 dBm, -140dBm (typ.)3.2 GHz~3.65 GHz-136 dBm, -143 dBm (typ.)-136 dBm, -143 dBm (typ.)3.65 GHz~4.1 GHz-136 dBm, -143 dBm (typ.)4.1 GHz~5.0 GHz-135 dBm, -141 dBm (typ.)5.0 GHz~5.85 GHz-135 dBm, -141 dBm (typ.)5.85 GHz~6.7 GHz-134 dBm, -140 dBm (typ.)6.7 GHz~7.5 GHz-132 dBm, -138 dBm (typ.)100 kHz ~1 MHz-125 dBm, -137 dBm (typ.)-125 dBm, -137 dBm (typ.)1 MHz~600 MHz-156 dBm, -163 dBm (typ.)-156 dBm, -163 dBm (typ.)600 MHz~1.8GHz-159 dBm, -165 dBm (typ.)-159 dBm, -165 dBm (typ.)1.8 GHz~3.2 GHz-157 dBm, -163 dBm (typ.)-157 dBm, -163 dBm (typ.)3.2 GHz~3.65 GHz-157 dBm, -163 dBm (typ.)-157 dBm, -163 dBm (typ.)3.65 GHz~4.1 GHz-157 dBm, -163 dBm (typ.)4.1 GHz~5.0 GHz-156 dBm, -162 dBm (typ.)5.0 GHz~5.85 GHz-156 dBm, -162 dBm (typ.)5.85 GHz~6.7 GHz-155 dBm, -161 dBm (typ.)6.7 GHz~7.5 GHz-153 dBm, -159 dBm (typ.)

SSB Phase Noise
Offset20 °C to 30 °C, fc = 1 GHz, Normalized to 1 Hz

10 kHz-100 dBc/Hz, -104 dBc/Hz (typ.)100 kHz-100 dBc/Hz, -104 dBc/Hz (typ.)1 MHz-114 dBc/Hz, -117 dBc/Hz (typ.)10

DataSheetFrequency Response
20 °C to 30 °C, 30% to 70% relative humidity, att = 20 dB, relative to 50 MHz
Preamp off±0.8 dB, ±0.4 dB (typ.)Preamp on±1.2 dB, ±0.6 dB (typ.)Error and Accuracy
Resolution bandwidth
switching uncertainty

Logarithmic resolution, relative to RBW = 10 kHz

Input attenuation
switching uncertainty

20 °C to 30 °C, fc = 50 MHz, preamp off, relative to att = 20 dB, att =0~50 dB± 0.2 dB (nom.)± 0.5 dB
20 °C to 30 °C, fc = 50 MHz, RBW= VBW = 1 kHz, att = 20 dB, peak detector, 95% reliability

Absolute amplitudeaccuracy

Total amplitude accuracy±0.4 dB, input signal -20 dBm, Preamp off±0.5 dB, input signal -40 dBm, Preamp on
20 °C to 30 °C, fc>100 kHz, input signal -50 dBm ~ 0 dBm, att = 20 dB, RBW=VBW=1 kHz,peak detector, preamp off, 95% reliability±0.7 dB
Att = 10 dB, fc≥1 MHz

RF input VSWR

1 MHz~3.2 GHz

1.7 (nom.)

3.2GHz~7.5 GHz

1.5 (nom.)

Distortion and Spurious Responses
20 °C to 30 °C, fc ≥ 50 MHz, mixer level -20 dBm, att = 0 dB, preamp off
Second harmonic distortion(SHI)

Third-order intercept(TOI)

1dB gain compression

Residual response
Input related spurious

DataSheet

50 MHz~3.2 GHz-65 dBc / +45 dBm (nom.)

3.2 GHz~3.75 GHz-80 dBc / +60 dBm (nom.)
20 °C to 30 °C, fc ≥ 50 MHz, two -20 dBm tones spaced by 100 kHz, att = 0 dB, preampoff
50 MHz~3.2 GHz+9.5 dBm (typ.)3.2 GHz~7.5 GHz+16 dBm (typ.)
20 °C to 30 °C, fc ≥ 50 MHz, two tones frequency interval ≥ 10MHz, RBW <1 kHz,att = 0
dB, preamp off> 8 dBm (nom.)
20 °C to 30 °C, input terminated = 50 Ω, att = 0 dB
< -90 dBm
20 °C to 30 °C, mixer level = -30 dBm<-65 dBc

11

Source (SHA850-SOR)
Frequency Parameter
SHA851A

SHA852A

Frequency Range

100 kHz ~ 3.6 GHz

100 kHz ~ 7.5 GHz

Frequency resolution

1 Hz

Source TypeCW，CW Offset

Power Parameter
Output level-40 dBm ~ 0 dBm

Output level resolution

1 dB

Output flatness±2 dB (nom.)

VSWR< 2 (nom.)

Connector and ImpendenceN-type female, 50 Ω

Average safe reverse power

Total 27 dBm (0.5 W)

Maximum safe reverse level±50 V DC

Advanced Measurement Kit (SHA850-AMK)
Power Measurement
CHP, Channel Power

Channel Power, Power Spectral Density

ACPR, Adjacent Channel Power
Ratio
OBW, Occupied Bandwidth

Main CH Power, Left channel power, Right channel power

T-Power, Time Domain Power

Zero Span Integrated Power

CNR, Carrier Noise Ratio

C/N, Noise Power

Non-Linear Measurement
Harmonic measurement

Max Harmonic number 10

TOI, Third-Order Intercept

Measure the third-order products from two tones

Occupied Bandwidth, Transmit Frequency Error

Spectrum Monitor Measurement
Spectrogram

12

DataSheetCable and Antenna Test
Measurement
SHA851A

SHA852A

Frequency Range

100 kHz~3.6 GHz

100 kHz~7.5 GHz

Sweep Points

101~10001, default 1001

Port1 Stimulus Power

Resolution (meters)-40dBm ~ 0dBm (nom.)(Sweep Points - 1) x Velocity Factor x Light of Speed (m/s) /(Stop Frequency - Start Frequency (Hz) )Maximum Distance / Sweep Points

Calibration

Open Response, Short Response, Response Through, Full 1-Port(OSL)

Velocity Factor

0.1~1

Cable Loss

-10 dB/m ~ 100 dB/m

Trace

Cable Loss(1-Port)

Mem, Math, Hold, Display
DTF,
Return Loss,
VSWR,
Cable Loss(1-Port),
Insertion Loss(2-Port),
TDR,
DTF & TDR, DTF & Return Loss, TDR & Return Loss
Locate problems or faults in a length of cable or transmission line
Format: Log Mag(dB), VSWR, Lin Mag
Distance Unit: Meters, Feet
Window Type: Rectangular, Hamming
Locate problems and identify the type of problem in a length of cable or transmission
line.
Format: Impedance(ohm), linear rho
Distance Unit: as DTF
Stimulus Type: Impulse, StepFrequency Type: Low-passWindow Type: Kaiser
Kaiser β: 0~13
Time Gate Type: Band Pass, Notch
Time Gate Shape: Normal, Maximum, Wide, Minimum
Time Gate Range: Start Distance ~ Stop Distance
Measure the accumulated losses throughout the length of the cable

Insertion Loss(2-Port)

Measure the loss through a DUT or cable over a specified frequency range

Maximum Distance (meters)

Meas TypeDistance to Fault(DTF)

Time Domain Reflectometry(TDR)

DataSheet13

Vector Network Analyzer (SHA850-VNA)
Stimulus and Measurement
SHA851A

SHA852A

Frequency Range

100 kHz ~ 3.6 GHz

100 kHz ~ 7.5 GHz

Measurement

S11, S21

IFBW

100 Hz，300 Hz，1 kHz，3 kHz，10 kHz

Port1 Stimulus Power

Sweep Points-40 dBm ~ 0 dBm (nom)
Lin Mag, Log Mag, Phase, Group Delay, SWR,
Smith Chart (Lin/Phase, Log/Phase, Real/Imag, R+j*X, G+j*B),
Polar Chart (Lin/Phase, Log/Phase, Real/Imag)
101~10001, default 1001

Trace

4 traces, Mem, Math, Hold, OverlayMarker(6+Ref)* 4 traces

FormatCalibration
Directivity of Calibration

F504MS, Log mag, Average=50, >50MHz> 40 dB (nom.)
S21, IFBW=10 kHz, Port1 level=0 dBm, Log Mag, Average=100

Dynamic
Range

100 kHz ~ 1 MHz

100 dB, 108 dB (typ.)100 dB, 108 dB (typ.)1 MHz ~ 1.5 GHz

108 dB, 114 dB (typ.)108 dB, 114 dB (typ.)1.5 GHz ~ 3.6 GHz

106 dB, 112 dB (typ.)106 dB, 112 dB (typ.)3.6 GHz ~ 6.5 GHz

102 dB, 109 dB (typ.)6.5 GHz ~ 7.5 GHz

100 dB, 107 dB (typ.)

Reflection trace noise (IFBW=10
kHz)

Transmission trace Noise (IFBW=10
kHz)

frequency

amplitude（dB rms）

phase（deg rms）

100 kHz~3.5 GHz

0.02

0.3

3.5 GHz~7.5 GHz

0.03

0.5

frequency

amplitude（dB rms）

phase（deg rms）

100 kHz~3.5 GHz

0.015

0.18

3.5 GHz~7.5 GHz

0.015

0.40

Short Response
Open Response
Calibration Type

Full 1-Port(OSL)
Response ThroughEnhanced Response

Port Extensions

Port 1, Port 2, Auto Open Port 1

System Z0

50 Ω

Velocity Factor

0.1~1

14

DataSheetAnalog Modulation Analysis (SHA850-AMA)
Common Parameter
SHA851A

SHA852A

Carrier Frequency Range

2 MHz ~ 3.6 GHz

2 MHz ~ 7.5 GHz

Carrier Power Accuracy±2 dB (nom.)

Carrier Power Range-30 dBm to +20 dBm (nom.)AM
Modulation rate range
Accuracy

20 Hz to 100 kHz
1 Hz (nom.)

Modulation rate < 1 kHz< 0.1% modulation rate (nom.)

Modulation rate ≥ 1 kHz

Modulation depth range

5% to 95%

Accuracy±4% (nom.)FM
Modulation rate range
Accuracy

20 Hz to 100 kHz
1 Hz (nom.)

Modulation rate < 1 kHz< 0.1% modulation rate (nom.)

Modulation rate ≥ 1 kHz

Frequency deviation

1 kHz to 400 kHz

Accuracy±4% (nom.)PM
Modulation rate range
Accuracy

50 Hz~50 kHz
1 Hz(nom.)

Modulation rate < 1 kHz< 0.1% modulation rate (nom.)

Modulation rate ≥ 1 kHz

Phase deviation

0.2~100 rad

Accuracy±4%(nom.)

DataSheet15

Digital Modulation Analysis (SHA850-DMA)
Common Parameter
SHA851A

SHA852A

Frequency Range

2 MHz ~ 3.6 GHz

2 MHz ~ 7.5 GHz

Carrier Power Accuracy± 2 dB (nom.)

Carrier Power Range-30 dBm to +20 dBm (nom.)

Measurement

Modulation TypeASK: 2ASK;
FSK: 2FSK, 4FSK, 8FSK, 16FSK;MSK: GMSK;PSK: BPSK, QPSK, OQPSK, 8PSK;DPSK: DBPSK, DQPSK, D8PSK, π/4 -DQPSK, π/8 -D8PSK;
QAM: 16, 32, 64, 128, 256

Meas Length

16 to 4096

Points/Symbol

4, 6, 8, 10, 12, 14, 16

Symbol Rate

1 ksps to 5 Msps, Symbol Rate* Points/Symbol ≤20 Msps

Trigger Holdoff

500 msBurst

Burst power sync, BERT

FilterMeas/Ref Filter

Nyquist, Squrt Nyquist, Gauss, Half Sine, Rectangular

Length2 to 128

Alpha/BT

Alpha 0.01~1, BT 0.01~10

Trace
Trace Data
Trace Formats

IQ Meas Time, IQ Meas Spectrum, IQ Ref Time, IQ Ref Spectrum,
Time, Spectrum, IQ Mag Err, IQ Phase Err
Symbol Error Chart, Err Vector Time, Err Vector Spectrum,Log mag, Lin mag, Real, Imag, I-Q, Constellation, I-eye, Q-eye,Wrap Phase, Unwrap Phase, Trellis eye

Symbol Error Chart
PSK/DPSK/MSK/QAM

EVM (rms EVM, peak EVM), Magnitude error,
Phase error, IQ offset, Carrier offset, SNR Quadrature error,
Gain imbalance(not support for MSK)ASKASK Error, ASK depth, carrier offset

FSK

FSK Error, Magnitude error, FSK deviation, carrier offset

16

DataSheetIndoor and outdoor map (SHA850-MAP)
Measurement
MAP Type(No built-in map)
MAP Level
Map Data

Indoor Map: PNG or JPEG
Outdoor Map: Google tile Format (Download from Google Maps)
Indoor Map: none
Outdoor Map: Display level range 1-19
Indoor Map: kdata recording and playback
Outdoor Map: kml recording and playback

GPSA Measurement
Channel Power measurement

IQ Data Acquisition (SHA850-IQA)
Measurement
Acquisition Setup

Acq Time、Sample Rate、Save(.txt)、Storage Device(Internal/USB)

Maximum Sample Rate

20 MHzMaximum Bandwidth

5 MHz

Data Sample Size

I=Q=2 Byte

Storage Depth

1 GB

Maximum Sample Length

250 MB(1 GB/4 Byte)

Sample Time Length

Sample Length/ Sample Rate

DataSheet

17Inputs and Outputs
Front Panel
RF input, Port 2

N-type female, 50 Ω (nom.)

Source, Port 1

N-type female, 50 Ω (nom.)

USB Host

USB-A plug, version 2.0

Ear Phone Jack

3.5 mmUSB Device

USB-C plug, version 2.0

LAN
GPS Antenna(SHA850-GPS)
Bias out(SHA850-BIAS)
10 MHz reference input

LAN (VXI 11), 10/100 Base,RJ-45

External trigger input

1 KΩ, 5V TTL level, BNC-type female

SMA(F), 3.3 V, 50 Ω
SMB(F), 12 V-32 V，0.1 V step
10 MHz, -5 to +10 dBm, BNC-type female, 50 Ω (nom.)

Remote Control
Communication Interface
Remote Control Capability

18

LAN, USB-TMC,GPIB (USB-GPIB adaptor)
SCPI / Labview / IVI based on USB-TMC / VXI-11 / GPIB / Socket / Telnet
NI-MAXWeb Browser (HTML 5 Supported)

DataSheetGeneral Specification
Structure
Dimensions

310 mm × 215 mm × 78.5 mm (W×H×D)

Weight

Net: 3.20 kg (7.0 lb)

Display

TFT LCD, 800 × 600, 8.4 inch multi-touch screen

Storage

Internal (Flash) 3.2 GByte, external (USB storage device) 32 GByte

Working Environment
SourceAC voltage range: 100-240 V, 50/60 Hz or 100-120 V, 400 Hz;

Power consumption

Altitude

20 W (typ.)Working temperature: 0 °C to 50 °C
Working while charging temperature: 0 °C to 45 °C
Storage temperature: -20 °C to 70 °C
0 °C to 30 °C, ≤ 95% Relative humidity
30 °C to 50 °C, ≤ 75% Relative humidity
Operating: less than 3 km (10000 feet)

Calibration cycle

1 yearTemperature
Humidity

Electromagnetic Compatibility
EN 61326-1: 2013 /Class A (The active input power of the EUT is less than 75 W. According to EN 61000-3-2, noEN 61000-3-2: 2014
limits are necessary.)
Plt: 0.65 Pst: 1.00, dmax: 4.00 % dc: 3.00 %EN 61000-3-3: 2013
dt Lim: 3.30 % dt>Lim: 500ms
SafetyCAN/CSA-C22.2 No. 61010-1:2012/A1:2018-11
CAN/CSA-C22.2 No. 61010-2-030:2018
UL 61010-1:2012/R:2018-11
UL 61010-2-030:2018
RoHS
2011/65/EU

DataSheet

19Ordering Information
Product

Product CodeStandard
Accessories

Options

GeneralAccessories

CAT&VNA
Accessories

20

Description

Order Number

Spectrum Analyzer 9 kHz~3.6 GHz,
Cable and Antenna Test 100 kHz~3.6 GHz

SHA851A

Spectrum Analyzer 9 kHz~7.5 GHz,
Cable and Antenna Test 100 kHz~7.5 GHz

SHA852A

Quick Start, USB type-C cable, Power cord, AC-DC adapter, Rechargeable
lithium battery, Portable bag
SHA851A to SHA852A

SHA850-F2

Source

SHA850-SOR

Vector Network Analysis

SHA850-VNA

Advanced Measurement Kit

SHA850-AMK

Analog Modulation Analysis

SHA850-AMA

Digital Modulation Analysis

SHA850-DMA

IQ Data Acquisition

SHA850-IQA

DC Bias Out

SHA850-BIAS

GPS Receiver

SHA850-GPS

Indoor and outdoor map

SHA850-MAP

Rechargeable lithium battery

10V8_BAT

AC-DC adapter

12V_AP_4A

Portable bag

BAG_H2GPS antenna, SMA(M), 1000 mm

ANT-GPS1

Directional Antenna Suit:
Wideband Antennas (10 MHz~200 MHz, 200 MHz~500 MHz,
500 MHz~8 GHz), Amplifier (10 dB, 9 kHz~8 GHz)

ANT-DA1

Near field probe kit:
H-field probes (20 mm,10 mm,5 mm), E-field probe (5 mm),
300 kHz~3 GHz

SRF5030T

Utility Kit:
N(M)-SMA(M) cable (6 GHz), N(M)-N(M) cable (6 GHz),
N(M)-BNC(F) adapter x2, N(M)-SMA(F) adapter x2,
10dB 1W attenuator

UKitSSA3X

N(M)-BNC(M) cable, DC~2 GHz, 700 mm

N-BNC-2L

N(M)-SMA(M) cable, DC~6 GHz, 700 mm

N-SMA-6L

N(M)-N(M) cable, DC~6 GHz, 700 mm

N-N-6L

N(M)-N(M) cable, DC~18 GHz, 1000 mm

N-N-18L

N(M)-SMA(M) cable, DC~18 GHz, 1000 mm

N-SMA-18L

SMA(M)-SMA(M) cable, DC~18 GHz, 1000 mm

SMA-SMA-18L

N type Integrated Precision Calibration Kit, Male, DC~9 GHz,50 Ω

Y504MSN type Integrated Precision Calibration Kit, Female, DC~9 GHz,50 Ω

Y504FSN type Precision Calibration Kit, DC~9 GHz, 50 Ω

F504TS

3.5mm type Precision Calibration Kit, DC~9 GHz, 50 Ω

F604TSN type Economic Calibration Kit, DC~4.5 GHz, 50 Ω

F503MEN type Economic Calibration Kit, DC~4.5 GHz, 50 Ω

F503FE

3.5mm type Economic Calibration Kit, DC~4.5 GHz, 50 Ω

F603ME

3.5mm type Economic Calibration Kit, DC~4.5 GHz, 50 Ω

F603FEN type Standard Calibration Kit, DC~9 GHz, 50 Ω

F504MS

DataSheetProduct

DataSheet

Description

Order Number

N type Standard Calibration Kit, DC~9 GHz, 50 Ω

F504FS

3.5mm type Standard Calibration Kit, DC~9 GHz, 50 Ω

F604MS

3.5mm type Standard Calibration Kit, DC~9 GHz, 50 Ω

F604FS

21