SHS800X
SHS1000X
Handheld Digital
Oscilloscope

Data Sheet
Rev. 01D
April. 2022

0
WWW.SIGLENT.COM

SHS820X/SHS1202X
SHS810X/SHS1102X

Product Overview
SIGLENT’s Handheld Digital Oscilloscopes include the SHS800X andSHS1000X models. Both feature 2 analog oscilloscope input channels andare available in 200 MHz and 100 MHz analog bandwidths, a single ADC
with 1 GSa/s maximum sample rate, and a single memory module with 12
Mpts memory depth. When two channels are enabled, each channel has
sample rate of 500 MSa/s and a standard record length of 6 Mpts. When
only a single channel is actived, the maximum sample rate is 1 GSa/s andthe maximum record length is 12 Mpts. For ease-of-use, the most commonlyused functions can be accessed with its user-friendly front panel design.
The SHS800X series has two non-isolated scope channels and one
isolated multimeter channel.
The SHS1000X series features full isolation between the two
oscilloscope channels, one multimeter channel, power adapter and the USB
host/device port. The full isolation makes it ideal for both laboratory andfloating signal measurement because it reduces the risk of accidental shortcircuits.
The max voltage input to the analog scope inputs is CATIII 600 Vrms,
CATII 1000 Vrms. And the max input for the multimeter is CATIII 600 Vrms,
CATII 1000 Vrms.
The SHS series utilizes a new generation of SPO (Super-Phosphor
Oscilloscope) technology that provides excellent signal fidelity andperformance. It comes with a vertical input range from 2 mV/div to 100 V/div,
and a max offset range up-to to 400 V.
The innovative digital trigger system delivers high sensitivity and low
jitter, and a waveform capture rate of up-to 400,000 frames/sec. The SHS
also employs a 256-level intensity grading display function and a color
temperature display mode for clarity and fast fault identification.
The SHS’ have multiple powerful triggering modes including serial bus
triggering as well as free decoding for IIC, SPI, UART, CAN, LIN bus types.
The SHS models also include History waveform recording, and sequential
triggering that enable extended waveform recording and analysis. Anotherpowerful addition is the new 1 million point FFT math function that gives the
SHS very high frequency resolution when observing signal spectra. The new
digital design also includes a hardware co-processor that delivers
measurements quickly and accurately without slowing acquisition and frontpanel response.
The SHS series feature built-in sample and measurement value
recorders. The sample recorder can achieve a maximum sampling rate of 25
kSa/s, and can record for up to 22 hours at this sampling rate. The
measurement value recorder can simultaneously record 4 sets ofmeasurement values at a recording interval as low as 0.1s which delivers
up-to 25 hours of continuous recording.
The SHS series integrates a 6000 count digital multimeter, isolated
from the scope that can measure DC/AC Voltage, DC/AC Current,
Resistance, Capacitance, Diodes, and Continuity.
A rechargeable 6900 mAh lithium battery supports up-to 5.5 hours (4hours SHS1000X) of work without a charge.
The handheld oscilloscopes feature a sealed IP51 dust and drip-proof
housing and has been tested to IEC60529 to guarantee the ruggedness thatis needed to survive harsh environments. A rubberized surface with large
keys also makes it easy to use in difficult environments.

WWW.SIGLENT.COM

Key Features
200 MHz, 100 MHz bandwidth models
Sample rate of 1 GSa/s (single-channel), Sample rate of
500 MSa/s (two-channels).
The Siglent SPO technology
 Waveform capture rates up to 100,000 wfm/s (normalmode) and 400,000 wfm/s (sequence mode) Supports 256-level intensity grading and color
temperature display modes Record length up to 12 Mpts
 Digital trigger system
Intelligent trigger: Edge, Slope, Pulse Width, Window, Runt,
Interval, Time out (Dropout), Pattern
Serial bus triggering and decoding (Standard) for IIC, SPI,UART, CAN, and LIN protocols
Video trigger/HDTV
Low background noise with voltage scales from 2 mV/div to
100 V/div
3 one-button shortcuts for Oscilloscope, Multimeter andRecorder functions
8 one-button shortcuts for: Run/Stop, Auto Setup, Default,
Measure, Cursors, Display/Persist, Clear Sweep and Print.
More function shortcuts available when combined with theshift button
Segmented acquisition (Sequence) mode, divides the
maximum record length into multiple segments (up to
80,000), according to trigger conditions set by the user,
with a very small dead time segment to capture thequalifying event
History waveform record (History) function (maximum
recorded waveform length is 80,000 frames)
Automatic measurement function for 38 parameters as well
as Measurement Statistics, Zoom, Gating, Math, History
and Reference functions
1 Mpts FFT. Support Peaks and Markers
Math and measurement functions use all sampled data
points (up to 12 Mpts)
Math functions (FFT, addition, subtraction, multiplication,
division, integration, differential, square root)
Default key can be customized for user settings or factory“defaults”
Supports Multi-language display and embedded online help
Security Erase mode
Search and navigate function
Includes Recorder mode, including Sample andMeasurement Loggers
6000 counts Digital Multimeter, Support DCV, ACV, DCI,
ACI, Resistance, Diode, Capacitance, Continuity test.
True RMS AC Voltage/Current measurement multimeter5.6-inch TFT-LCD display with 640 * 480 resolution
Interface types: Isolated USB Host, USB Device (MicroUSB-TMC)
Supports SCPI remote control commands
UL2054 certified lithium battery pack, 6900 mAh capacity,
external charger
IP Rating: IP51
Compliance with UL61010-1, UL61010-2-030, UL61010-2033

1

SHS800X&SHS1000X Series Handheld Digital Oscilloscope

Models and Key Specifications
Model
Bandwidth
Sample rate
(Max.)
Channels
Memory depth(Max.)
Waveform
capture rate
(Max.)
Trigger type
Serial Trigger anddecoder
Data
Logger(Recorder)
I/O
Max input
Voltage(Scope)Max input
Voltage(Meter)
Probe
Display
Weight2

SHS810X
SHS820X
SHS1102X
SHS1202X
100 MHz
200 MHz
100 MHz
200 MHz
Two-channel share a single 1 GSa/s ADC. When two channels are enabled, each channel has
a maximum sample rate of 500 MSa/s. When a single channel is active, that channel has asample rate of 1 GSa/s
2 analog oscilloscope channels, 1 multimeter channel
6 Mpts/CH (dual-channel mode)
12 Mpts/CH (single channel mode)
100,000 wfm/s (normal mode), 400,000 wfm/s (sequence mode)
Edge, Slope, Pulse Width, Window, Runt, Interval, Dropout, Pattern, Video
IIC, SPI, UART, CAN, LIN
Sample Logger. The Max sample rate is 25 kSa/s, the Min sample rate is 1 Sa/s
Measurement Logger. The Max interval is 10 minutes, the Min interval is 0.1second. The Max
items of logging is 4
USB Host, USB Device
CAT II 300 Vrms Between BNC Signal and CAT III 600 Vrms, CAT II 1000 Vrms Between
Protecting Earth
BNC Signal and Protecting EarthCAT II 30 Vrms Between BNC GND and CAT III 600 Vrms, CAT II 1000 Vrms Between
Protecting EarthBNC GND and Protecting EarthCAT II 300 Vrms Between BNC Signal and CAT III 300 Vrms Between BNC Signal and
BNC GNDBNC GND
CAT III 300 Vrms, CAT II 600 Vrms

CAT III 600 Vrms, CAT II 1000 Vrms

PP510
PP215
5.6-inch TFT-LCD (640x480)
Without package 1.75 kg. With package 3.5 kg

PB925

WWW.SIGLENT.COM

SHS800X&SHS1000X Series Handheld Digital Oscilloscope

Functions & Characteristics
Front panel and back panel

Front panel of the SHS800X series

Rear of the SHS800X series

Large bright 5.6-inch TFT -LCD display with 640 * 480 resolution. The most commonly used functions are accessible using 8
different one-button operation keys: Run/Stop, Auto Setup, Default, Cursor, Measure, Display/Persist, Clear Sweep, and Print.
More function shortcuts are available combined with the shift button.
When two channels are enabled, each channel has a maximum sample rate of 500 MSa/s. When a single channel is active,that channel has a sample rate of 1 GSa/s

Record Length of up to 12 Mpts

Using hardware-based Zoom technologies and max record length up to
12 Mpts, users can oversample to capture for longer periods at higher
resolution and use the zoom feature to see more details within eachsignal.

WWW.SIGLENT.COM

Waveform Capture Rate up to 400,000 wfm/s

With a waveform capture rate of up to 400,000 wfm/s (sequencemode), the oscilloscope can easily capture unusual or low-probability
events.

3

SHS800X&SHS1000X Series Handheld Digital Oscilloscope

256-Level Intensity Grading and Color Temperature Display

SPO display technology provides fast refresh rates. The resulting
intensity-graded trace is brighter for events that occur with more
frequency and dims when the events occur with less frequency.

Serial Bus Decoding Function

The color temperature display is similar to the intensity-graded trace
function, but the trace occurrence is represented by different colors(color “temperature”) as opposed to changes in the intensity of one
color. Red colors represent events that occur more frequently, while blue
is used to mark points that occur less frequently.

History Waveforms (History) Mode and Segmented
Acquisition (Sequence)

SHS800X/SHS1000X displays the decoding through the events list. Bus
protocol information can be quickly and intuitively displayed in a tabularformat.

True measurement to 12 M points

SHS800X/SHS1000X series can measure all sampled data points up to
12 Mpts. This ensures the accuracy of measurements while the mathco-processor decreases measurement time and increases ease-of-use.

4

Playback the latest triggered events using the history function.
Segmented memory collection will store trigger events into multiple (Upto 80,000) memory segments, each segment will store triggered
waveforms and timestamps for each frame.

Gate and Zoom Measurement

Through Gate and Zoom measurement, the user can specify anarbitrary interval of waveform data analysis and statistics. This helps
avoid measurement errors that can be caused by invalid or extraneous
data, greatly enhancing the measurements’ validity and flexibility.

WWW.SIGLENT.COM

SHS800X&SHS1000X Series Handheld Digital Oscilloscope

1M points used to calculate the FFT

The new math co-processor enables FFT analysis of incoming signals
using up to 1 M samples per waveform. This provides high-frequency
resolution with a fast refresh rate. The FFT function also supports avariety of window functions so that it can adapt to different spectrum
measurement needs. Support Peaks, Markers, a variety of numbers.

Search and Navigate

The SHS800X/SHS1000X series can search events specified by theuser in a frame. It can also navigate by time (delay position) andhistorical frames.

Sample Logger

The Sample Logger is the mode of logging the sampling points for along time. For there are many sampling points to log, they are logged
into the internal flash or external U disk in real-time. After stopping
logging, the user can recall the sampling points on the oscilloscope, or
analyze the saved data on the computer.

WWW.SIGLENT.COM

Customizable Default Key

The current parameters of the oscilloscope can be preset to Default Key
through the Save menu.

6000 Counts Digital Multimeter

6000 count digital multimeter featured function of DCV, true RMS ACV,
DCI, ACI, Diode, Resistance, Capacitance, and Continuity.

Measurement Logger

The measurement Logger is the mode of logging the measurement
value for a long time. For the amount of measurement data is relatively
small, to process quickly, the data is logged in memory. After stopping
logging, the data can be saved into the internal flash or external U disk.

5

SHS800X&SHS1000X Series Handheld Digital Oscilloscope

Adapter/Battery

Wall power using the supplied adapter
SHS800X/SHS1000X supports adapter power supply and battery
power supply. After connecting the adapter, the battery enters into
charging mode. The adapter provides a maximum 4 A output current.

Battery powered
SHS800X/SHS1000X uses a UL2054 certified lithium battery package.
The battery capacity of 6900 mAh can guarantee long-term operation
without an external power supply for up-to 5.5 hours (SHS800X) and 4
hours (SHS1000X). The battery supports an external charger to further
meet the requirements of portability.

Connectivity

Right side of the SHS800X series

Left side of the SHS800X series

SHS800X/SHS1000X supports USB Host, USB Device (Micro USB -TMC).

6

WWW.SIGLENT.COM

SHS800X&SHS1000X Series Handheld Digital Oscilloscope

Specifications
Oscilloscope
Acquisition System
Model
Sampling Rate (Max.)

SHS800X
1 GSa/s (single channel), 500 MSa/s (two channels)

Memory Depth (Max.)
Peak Detect
Average
ERES

Max 12 Mpts/Ch (single channel), 6 Mpts/Ch (two channels)
2 ns
Averages:4,16, 32,64,128,256,512,1024
Enhance bits:0.5, 1.5, 2, 2.5, 3

Waveform interpolationSin(x)/x, Linear

Input
Model
Channels
Coupling
Impedance

SHS800X
2 channels
DC, AC, GND
DC: (1 MΩ±2%) || (14 pF ±2 pF)

Max. Input voltage[1]CAT II 300 Vrms Between BNC Signal and Protecting
Earth
CAT II 30 Vrms Between BNC GND and Protecting EarthCAT II 300 Vrms Between BNC Signal and BNC GND

CH to CH Isolation

DC-Max BW: >40 dB

SHS1000X

SHS1000X

DC: (1 MΩ±2%) || (14 pF ±2 pF)
CAT III 600 Vrms, CAT II 1000 Vrms Between BNC
Signal and Protecting EarthCAT III 600 Vrms, CAT II 1000 Vrms Between BNC
GND and Protecting EarthCAT III 300 Vrms Between BNC Signal and BNC GND

Probe attenuation
0.1X,0.2X,0.5X,1X,2X,5X,10X……1000X,2000X,5000X,10000X,Custom
Notes [1]: According to IEC61010-1, a voltage higher than 30 Vrms is a dangerous voltage, necessary protection must be taken to prevent personal
injury. Please read the user’s manual for details.
Vertical System
Model
[2]

Bandwidth (-3 dB)

Vertical Resolution
Vertical Scale (Probe 1X)
Offset Range (Probe 1X)
Bandwidth limit[2]SHS800X≥ 200 MHz (SHS820X)≥ 100 MHz (SHS810X)
8-bit
2 mV/div–100 V/div (1-2-5 sequence )2 mV- 296 mV: ± 5 V
302 mV- 7.5 V: ± 80 V
7.6 V- 100 V: ± 400 V

DC- 10% (BW): ± 1 dB
10%- 50% (BW): ± 2 dB
50%- 100% (BW): + 2 dB/-3 dB

Low-frequency response(AC coupling -3 dB)≤ 2 Hz (at input BNC)SFDR including harmonics

2 mV/div: > 24 dB
5 mV/div: >25 dB≥ 10 mV/div: > 35 dB
P-P Noise <= 15 SDEV Spec
≥ 30 dB

CMRR
DC Gain Accuracy
Offset Accuracy
Rise time[2]Overshoot (500 ps Pulse) [2]

5 mV/div–100 V/div (1-2-5 sequence )20 MHz ± 40%

Bandwidth Flatness[2]Noise/SNRSHS1000X≥ 200 MHz (SHS1202X)≥ 100 MHz (SHS1102X)≥ 28 dB> 100 dB DC> 50 dB to AC 1 MHz≤ ± 3%: ≥ 10 mV/div≤ ± 4%: < 10 mV/div± (1.5%* Offset+1.5%*8*div+5 mV)
Typical 1.7 ns (SHS820X)
Typical 3.5 ns (SHS810X)±(1.5%* Offset+1.5%*8*div+5 mV)
Typical 2.0 ns (SHS1202X)
Typical 3.5 ns (SHS1102X)

typical 12%

typical 18%

Notes[2]: The SHS series handheld oscilloscope featured 1 MΩ input impedance. Bandwidth and pulse response must be verified with an external
50 Ω adapter, to guarantee signal integrity at higher frequency.
Horizontal System
Timebase Scale
Channel Skew
Waveform Capture Rate
Intensity grading
Display Format
Timebase Accuracy

1.0 ns/div-100 s/div< 300 ps
Up to 100,000 wfm/s (normal mode), 400,000 wfm/s (sequence mode)
256 Levels
Y -T, X -Y, Roll
±25 ppm

Roll Mode

50 ms/div-100 s/div (1-2-5 sequence)

WWW.SIGLENT.COM

7

SHS800X&SHS1000X Series Handheld Digital Oscilloscope

Trigger System
Mode
Level
Hold off range

Coupling

Coupling Frequency
Response
Accuracy (typical)
Sensitivity
Jitter
Displacement
Edge Trigger
Slope
SourceSlope Trigger
Slope
Limit Range

Auto, Normal, Single
Internal: ±4.5 div from the center of the screen
80 ns- 1.5 s
AC
DC
LFRJ
HFRJ
Noise RJ
DC: Passes all components of the signal
AC: Blocks DC components and attenuates signals below 8 Hz
LFRJ: Blocks the DC component and attenuates the low-frequency
components below 2 MHz
HFRJ: Attenuates the high-frequency components above 1.2 MHz
Internal: ± 0.2 div
DC - Max BW: 0.8 div< 100 ps
Pre-Trigger: 0 - 100% Memory
Delay Trigger: 0 to 10,000 div
Rising, Falling, Rising & Falling
All channels
Rising, Falling＜, ＞, ＜＞, ＞＜

SourceTime Range
Resolution

All channels
2 ns - 4.2 s
1 ns

Pulse Width Trigger
Polarity
Limit Range+wid , -wid

SourcePulse Range
Resolution

All channels
2 ns - 4.2 s
1 ns

Video Trigger
Signal Standard
Source＜, ＞, ＜＞, ＞＜

Sync
Trigger condition

NTSC, PAL, 720p/50, 720p/60, 1080p/50,1080p/60,1080i/50,1080i/60, Custom
All channels
Any, SelectLine, Field

Window Trigger
Window Type
SourceAbsolute, Relative
All channels

Interval Trigger
Slope
Limit Range

Rising, Falling

SourceTime Range
Resolution

All channels
2 ns - 4.2 s
1 ns

Dropout Trigger
Timeout Type
SourceSlope
Time Range
Resolution

Edge, StateAll channels
Rising, Falling
2 ns - 4.2 s
1 ns＜, ＞, ＜＞, ＞＜

Runt Trigger
Polarity
Limit Range+wid , -wid

SourceTime Range
Resolution

All channels
2 ns - 4.2 s
1 ns

Pattern Trigger
Pattern Setting
Logic
SourceInvalid, Low, HighAND, OR, NAND, NOR
All channels

8＜, ＞, ＜＞, ＞＜

WWW.SIGLENT.COM

SHS800X&SHS1000X Series Handheld Digital Oscilloscope

Limit Range＜, ＞, ＜＞, ＞＜Time Range
Resolution

2 ns - 4.2 s
1 ns

Serial Trigger
I2C Trigger
Condition
Source(SDA/SCL)
Data format
Limit Range
Data Length

R/W bitSPI Trigger
Condition
Source(CS/CL/Data)
Data format
Data Length
Bit Value
Bit Order
UART Trigger
Condition
Source(RX/TX)
Data format
Limit Range
Data Length
Data Width
Parity Check
Stop Bit
Idle Level
Baud Rate(Selectable)
Baud Rate (Custom)
CAN Trigger
Condition
SourceID
Data Format
Data Length
Baud Rate
LIN Trigger
Condition
SourceID
Data Format
Data Length
Baud Rate(Selectable)
Baud Rate (Custom)
SearchEvent
Event Number

Serial Decoder
Decoders
I2C
SignalAddress
Threshold
List

Start, Stop, Restart, No Ack, EEPROM, 7-bits Address & Data, 10-bits Address & Data, Data Length
All channels
Hex
EEPROM: =, >, <EEPROM: 1 byte
Addr & Data: 1-2 byte
Data Length: 1-12 byte
Addr & Data: Read, Write, Do not care
Data
All channels
Binary4-96-bit
0, 1, X
LSB, MSB
Start, Stop, Data, Parity Error
All channels
Hex=, >, <
1 byte5, 6, 7, 8-bits
None, Odd, Even, Space, Mark1, 1.5, 2-bits
High, Low
600/1200/2400/4800/960019200/38400/57600/115200/Custom bit/s
300-5000000 bit/s
Start, Remote, ID, ID + Data, Error
All channels
STD (11-bits), EXT (29-bit)Hex
1 -2 byte
5k/10k/20k/50k/100k/125k/250k/500k/800k/1M/Custom bit/s
Break, Frame ID, ID+Data, Error
All channels
1 byteHex
1-2 byte
600/1200/2400/4800/9600/19200/Custom bit/s
300 bit/s -20 Mbit/s

Edge, Slope, Pulse, Interval, Runt
Y-T: 600
ROLL: No limitation
Stop After ROLL: 600

2
SCL, SDA
7, 10 bits
-4.5 - 4.5 div
1- 7 lines

SPI

WWW.SIGLENT.COM

9

SHS800X&SHS1000X Series Handheld Digital Oscilloscope

SignalEdge SelectIdle Level
Bit Order
Threshold
List

SCL, MISO, MOSI, CS (2 channel scopes can only use 2 signal) identifiers
Rising, Falling
Low, High
MSB, LSB-4.5 - 4.5 div
1- 7 lines

UART
SignalData Width
Parity Check
Stop Bit
Idle Level
Threshold
List

RX, TX
5,6,7,8 bits
None, Odd, Even, Space, Mark1, 1.5, 2 bits
Low, High
-4.5 - 4.5 div
1- 7 lines

CAN
SignalSourceThreshold
List

CAN_H, CAN_L
CAN_H, CAN_L, CAN_H-CAN_L-4.5 - 4.5 div
1- 7 lines

LIN
LIN Specification Package
Revision
Threshold
List
Measurement
SourceNumber of
Measurements
Measurement Range
Measurement Parameters

VerticalVer1.3, Ver2.0-4.5 - 4.5 div
1- 7 lines

All channels, All channels in Zoom, Math, All References, History
Display 4 measurements at the same time. 5 measurements are displayed in the statistics table.
Screen or Gate region
38 Types
Max
Min
Pk-Pk
Ampl
Top
Base
Mean
Cmean
Stdev
Cstd
VRMS
Crms
FOV
FPRE
ROV
RPRE
Level@X
Period
Freq
+Wid
-Wid
Rise Time
Fall Time
Bwid

Horizontal
+Duty
-Duty
Delay
Time@Level

10

Highest value in input waveform
The lowest value of the input waveform
Difference between maximum and minimum data values
Difference between top and base in a bimodal signal, or between max and min in a
unimodal signal
Value of most probable higher state in a bimodal waveform
Value of most probable lower state in a bimodal waveform
Average of all data values
Average of data values in the first cycle
Standard deviation of all data values
Standard deviation of all data values in the first cycle
Root mean square of all data values
Root mean square of all data values in the first cycle
Overshoot after a falling edge;(base -min)/Amplitude
Overshoot before a falling edge;(max -top)/Amplitude
Overshoot after a rising edge;(max -top)/Amplitude
Overshoot before a rising edge;(base -min)/Amplitude
the voltage value of the trigger point
Time between the middle threshold points of two consecutive, like-polarity edges
Reciprocal of period
Width measured at 50% level and positive slope
Width measured at 50% level and negative slope
Duration of rising edge from 10 -90%
Duration of falling edge from 90 -10%
Time from the first rising edge to the last falling edge, or the first falling edge to the lastrising edge at the 50% crossing
Time difference between the 50% threshold of a rising edge to the 50% threshold of thenext falling edge of the pulse
Time difference between the 50% threshold of a falling edge to the 50% threshold of thenext rising edge of the pulse
Time from the trigger to the first transition at the 50% crossing
Time from the trigger to each rising edge at the 50% crossing.
When Statistics is Off, it shows the time from the trigger to the last rising edge at the 50%
crossing.

WWW.SIGLENT.COM

SHS800X&SHS1000X Series Handheld Digital Oscilloscope

Statistics

When Statistics is On, it shows the Mean, Min, Max, Standard Deviation of time from thetrigger to each rising edge at the 50% crossing in multiple frames (number = Count). The
Current shows the time of the current frame from the trigger to the last rising edge at the
50% crossing.
Phase
Phase difference between two edges
FRFR
Time from the first rising edge of channel A to the following first rising edge of channel B
FRFF
Time from the first rising edge of channel A to the following first falling edge of channel B
FFFRTime from the first falling edge of channel A to the following first rising edge of channel B
FFFF
Time from the first falling edge of channel A to the following first falling edge of channel B
FRLRTime from the first rising edge of channel A to the last rising edge of channel B
FRLF
Time from the first rising edge of channel A to the last falling edge of channel B
FFLRTime from the first falling edge of channel A to the last rising edge of channel B
FFLF
Time from the first falling edge of channel A to the last falling edge of channel B
Skew
Time of source A edge minus time of nearest source B edge
Manual: Time X1, X2, (X1 -X2), (1/ΔT)
Voltage Y1, Y2, (Y1 -Y2)
Track: Time X1, X2, (X1 -X2)
Current, Mean, Min, Max, Stdev, Count

Counter

Hardware 6-digit counter（channels are selectable）

Math
Operation
FFT window+ , - , * , / , FFT , d/dt , ∫dt , √Rectangular, Blackman, Hanning, Hamming, Flattop

FFT display

Full Screen, Split, Exclusive

Delay

Cursors

RecorderSample Logger
SourceSample Rate
Memory Depth
Log Time with Max sample
rate
Data Format
Measurement Logger
SourceLog Interval
Number of simultaneous
logging channels
Memory Depth
Log Time with MinimumInterval
Data Format
Export Data Format

CH1, CH2, CH1 & CH2
1 Sa/s- 25 kSa/s (1-2-5 sequence)
Internal memory 50 MB, Support External memory to 2 GB
Approx. 23 mins in single-channel mode,11 mins in two channels mode with internal memory
Approx. 22 hours in single-channel mode,11 hours in two-channel mode with external memory
BinaryMeasurement, Meter, Measurement & Meter
0.1 s- 10min4
Approx.3.6 Msamples in single-channel mode, 900 ksamples in four-channel mode
Approx.100 hours
BinaryBinary, csv, MATLAB

Multimeter (DMM)【1】Maximum Resolution
Maximum Input Voltage(SHS800X)
Maximum Input Voltage(SHS1000X)
Maximum Input Voltage(For adapter SCD10A,
SCD600MA)
Function

DC Voltage

AC Voltage(45 Hz～400 Hz)

WWW.SIGLENT.COM

6000 Counts
CAT III 300 Vrms
CAT II 600 Vrms
CAT III 600 Vrms
CAT II 1000 Vrms
CAT III 60 Vrms

Range
60.00 mV
600.0 mV
6.000 V
60.00 V
600.0 V
1000 V[3]60.00 mV
600.0 mV
6.000 V
60.00 V
600.0 V
750 V[3]Resolution
10 uV
100 uV
1 mV
10 mV
100 mV1V
10 uV
100 uV
1 mV
10 mV
100 mV
1V

Accuracy[4](± 1% ± 15 digit)(± 1% ± 5 digit)(± 1.5% ± 5 digit)(± 1% ± 15 digit)( ±1% ± 5 digit)(± 1.5% ± 5 digit)

11

SHS800X&SHS1000X Series Handheld Digital Oscilloscope

DC Current[2] [5]AC Current[2] [5](45Hz～400Hz)

Resistance

CapacitanceDiode60.00 mA
600.0 mA
6.000 A
10.00 A

10 uA
100 uA
1 mA
10 mA

60.00 mA
600.0mA
6.000 A
10.00 A
600.0 Ω
6.000 kΩ60.00 kΩ
600.0 kΩ
6.000 MΩ
60.00 MΩ
40.00 nF
400.0 nF
4.000 uF
40.00 uF
400.0 uF

10 uA
100 uA
1 mA
10 mA
0.1 Ω
1Ω
10 Ω
100 Ω
1 kΩ
10 kΩ
0.01 nF
0.1 nF
1 nF
10 nF
100 nF(± 4% ± 10 digit)(± 5% ± 5 digit)(± 4 % ± 10 digit)(± 5% ± 5 digit)(± 1% ± 5 digit)(± 4% ± 5 digit)(± 5% ± 50 digit)(± 5% ± 5 digit)

0～2 V

Continuity
Continuous beep when resistance < 50 ΩNote: [1] The spec for DMM functions are calibrated and verified in Battery-Power mode, Temperature range [23℃ ± 5℃], warm-up for 0.5 hour.
Note: [2] For rank A (ampere) range, the measurement time should be less than 10s, the interval time should be more than 15 minutes.
Note: [3] This spec is for SHS1000X only, The maximum input voltage is 600V (DC/AC) for the SHS800X series.
Note: [4] ± of reading % ± range error. For AC signals, the input signal should be greater than 10% of range.
Note: [5] 60mA, 600mA specification along with adapter SCD600MA; 6 A, 10 A specification along with adapter SCD10A.
I/O
USB Host
USB device
Probe compensation
output1 port, isolated type A plug, Full/Low speed, memory sticks only
1 port, Micro USB-B, remote control only
1 kHz, 0~5 V Square wave output

Display (Screen)
Display Type
Display Resolution
Display Color
Contrast(Typical)

5.6–inch TFT LCD
640×480 pixels
24-bit
500:1

Backlight

200 nits

Display (Waveform)
Range
Display Mode
Persist Time
Color Display
Screen Saver

8 x 12 divisions
Dot, Vector
Off, 1 Sec, 5 Sec, 10 Sec, 30 Sec, Infinite
Normal, Color
1 min, 5 min, 10 min, 30 min, 1 hour, Off

Language

Simplified Chinese, Traditional Chinese, English, French, Japanese, Korean, German, Spain, Russian, Italian,Portuguese

Environmental
Temperature
Humidity
HeightStandards
Electromagnetic
compatibility

12

Operating: 0℃ - +40℃
Non-operating: -20℃ - + 60℃
Operating: 85% RH, 40 ℃, 24 hours
Non-operating: 85% RH, 65 ℃, 24 hours
Operating: ≤ 2000 m
Non-operating: ≤ 5000 m

Meets EMC directive (2014/30/EU), meets or exceeds IEC 61326-1:2012/EN61326-1:2013 (Basic)
Conducted disturbanceRadiated disturbance

CISPR 11/EN 55011
CISPR 11/EN 55011

CLASS A group 1,150 kHz-30 MHz
CLASS A group 1, 30 MHz-1 GHz

Electrostatic discharge (ESD)

IEC 61000-4-2/EN 61000-4-2

4.0 kV（Contact）,8.0 kV（Air）

Radio-frequency
field Immunity

IEC 61000-4-3/EN 61000-4-3

electromagnetic

10 V/m（80 MHz to 1 GHz）;
3 V/m（1.4 GHz to 2 GHz）;

WWW.SIGLENT.COM

SHS800X&SHS1000X Series Handheld Digital Oscilloscope

1 V/m（2.0 GHz to 2.7 GHz）
2 kV （Input AC Power Ports）

Electrical fast transients (EFT)IEC 61000-4-4/EN 61000-4-4

SurgesIEC 61000-4-5/EN 61000-4-5

Radio-frequency continuous
conducted Immunity

IEC 61000-4-6/EN 61000-4-6

3 V, 0.15-80 MHz

IEC 61000-4-11/EN 61000-4-11

Voltage Dips:
0% UT during 1 cycle
40% UT during 10/12 cycles
70% UT during 25/30 cycles
Voltage interruptions:0% UT during
250/300 cycles

Voltage dips and interruptions

1 kV （Line to line）

SafetyUL 61010-1:2012/R:2018-11; CAN/CSA-C22.2 No. 61010-1:2012/A1:2018-11.
UL 61010-2-030:2018; CAN/CSA-C22.2 No. 61010-2-030:2018.
UL 61010-2-033:2020.

Power Supply/Battery
ModelPower Adapter

SHS800X

SHS1000X

Input

100 ～ 240 Vrms 50/60 Hz, 1.2 A

100 ～ 240 Vrms 50/60 Hz, 1.1 A

OutputBattery
Operating time
Charging time
Capacity
Charging Protection
Power Consumption

9 V, 4 A

12 V, 4 A

5.5 hours
4 hours while the instrument is switched off
6900 mAh≥ 55℃ at Battery

4 hours
4 hours while the instrument is switched off

Battery Mode

9W

11 W

Mechanical
IP Rating

IP51
Length: 276 mm

Dimensions

Width: 168 mm
Height(Depth): 68 mm

Weight with Battery

WWW.SIGLENT.COM

Without package 1.75 Kg, With package 3.5 Kg

13

SHS800X&SHS1000X Series Handheld Digital Oscilloscope

Probes and Accessories
Probe

PictureModel

Specifications &Description
Bandwidth: 100 MHz,

PP510
1X/10X, 1 M/10 Mohm,1X CATII 150 V, 10X CATII 300 V
Passive
Bandwidth: 200 MHz,PP215
1X/10X, 1 M/10 Mohm,1X CATII 150 V, 10X CATII 300 V

High Voltage

Bandwidth: 250 MHz,
PB925

Passive

10X Fixed, 10 Mohm, CATIII 600 V, CATII 1000 V

Multimeter Test

Wire Length: 1 m,
SHS-MTL

Leads

Rated Voltage: CATII 1000 V

Bandwidth: 200 KHz,
Max. continuous current: 20 Arms Peak current: 60 A
CP4020Switch Ratio: 50 mV/A, 5 mV/A,
Accuracy: 50 mV/A (0.4 A -10 Apk) ± 2%, 5 mV/A (1 A-60 Apk) ± 2%
9 V battery source
Bandwidth: 1 MHz,
Max. continuous current: 50 Arms, Peak current: 140 A
CP4050Switch Ratio: 500 mV/A, 50 mV/A
Accuracy: 500 mV/A (20 mA -14 ApK) ± 3% ± 20 mA , 50 mV/A (200mA -100 ApK) ± 4% ± 200 mA, 50 mV/A (100 A -140 ApK) ± 15%max
9 V battery source
Current Probe

Bandwidth: 300 kHz,
Max. continuous current: 70 Arms, Peak current: 200 A
CP4070Switch Ratio: 50 mV/A, 5 mV/A, Accuracy: 50 mV/A (0.4 A -10 ApK)
±2% , 5 mV/A (1 A -200 ApK) ± 2%
9 V battery source
Bandwidth: 50 MHz,
CP5030Max. continuous current: 30 Arms, Peak current: 50 A
Switch Ratio: 100 mV/A, 1 V/A, Accuracy: 1 V/A (± 1% ± 1 mA), 100mV/A (± 1% ± 10 mA), DC 12 V/1.2 A power adapter
Bandwidth: 100 MHz,

CP5030A

Max. continuous current: 30 Arms, Peak current: 50 A
Switch Ratio: 100 mV/A, 1 V/A, Accuracy: 1 V/A (±1%±1 mA), 100mV/A (± 1% ± 10 mA), DC 12 V/1.2 A power adapter

14

WWW.SIGLENT.COM

SHS800X&SHS1000X Series Handheld Digital Oscilloscope

Bandwidth: 12 MHz,
CP5150Max. continuous current: 150 Arms, Peak current: 300 A
Switch Ratio: 100 mV/A, 10 mV/A, Accuracy: 100 mV/A (± 1% ± 10mA), 10 mV/A (± 1% ± 100 mA), DC 12 V/1.2 A power adapter
Bandwidth: 5 MHz,

CP5500Max. continuous current: 500 Arms, Peak current: 750 A
Switch Ratio: 100 mV/A, 10 mV/A, Accuracy: 100 mV/A (± 1% ± 10mA), 10 mV/A(±1% ± 100 mA), DC 12 V/1.2 A power adapter

Bandwidth: 50 MHz,DPB4080
Differential Range: 800 V (DC + Peak AC),
100X/200X/500X/1000X, Accuracy: ±1%, DC 9 V/1 A power adapter

Bandwidth: 70 MHz,DPB5150
Differential Range: 1500 V (DC + Peak AC), 50X/500X
Accuracy: ±2 %, DC 5 V/1 A USB adapter

Bandwidth: 100 MHz,DPB5150A
Differential Probe

Differential Range: 1500 V (DC + Peak AC),
50X/500X , Accuracy: ± 2%
DC 5 V/1 A USB adapter
Bandwidth: 70 MHz,DPB5700

Differential Range: 7000 V (DC + Peak AC),
100X/1000X , Accuracy: ± 2%,
DC 5 V/1 A USB adapter
Bandwidth: 100 MHz
Differential Range: 7000 V (DC + Peak AC),

DPB5700A
100X/1000X
Accuracy: ± 2%
DC 5 V/1 A USB adapter
Bandwidth: 40 MHz
Differential Range: DC 10 kV, AC (rms): 7 kV (sine), AC (Vpp): 20 kV
HPB4010High Voltage(Pulse)1000X
Accuracy: ≤ 3%
Provides isolation between standard oscilloscope channels, isolation
ISFE

between the measured signal and ground.

Isolated front end
Uses USB 5 V power supply, plug and play.
The maximum input voltage allowed is up to ± 600 Vpk.

WWW.SIGLENT.COM

15

SHS800X&SHS1000X Series Handheld Digital Oscilloscope

STB-3 Test
Output signals including square, sine, AM, fast edge, pulse, PWM, I2C,
Demo Board

Board
CAN, LIN etc. Used in teaching and demonstrations.

Sampling Resistance: 0.5 Ω,
SCD600MA
Max. Working Current: 600 mA.

Current Adapter

Sampling Resistance: 5 mΩ,SCD10A
Max. Working Current: 10 A,10 sec.

Input Voltage: 9-26 V DC, Output current: 0-4 A,
Smart Battery

Max output power:40 W, Charging efficiency:>85%,
GSCH4000A

Charger

Dimensions(L x W x H): 180 mm*92 mm*58 mm,
Weight without battery: 235 g.

Color: Black,
Carrying Bag

BAG-H1

Dimensions(L x W x H): 325 mm*310 mm*95 mm,
Weight: 200 g.

16

WWW.SIGLENT.COM

SHS800X&SHS1000X Series Handheld Digital Oscilloscope

Ordering Information
Ordering information
Product Name

Standard Accessories

Optional Accessories

SHS820X 200 MHz
SHS810X 100 MHz
SHS1202X 200 MHz Isolated Input
SHS1102X 100 MHz Isolated Input
USB Cable -1
Quick Start -1
Passive Probe -2
Multimeter Test Lead -2
Certification -1
Power Adapter -1
Battery -1
SCD600MA Current Measurement Adapter -1
SCD10A Current Measurement Adapter -1
Carrying Bag -1
Isolated Front End
STB Demo Source
High Voltage Probe
Current Probes
Differential Probes
Smart Battery Charger

WWW.SIGLENT.COM

ISFE
STB-3
HPB4010CP4020/CP4050/CP4070/
CP4070A/CP5030/CP5030A/CP5150/CP5500/CPL5100
DPB1300/DPB4080/DPB5150/DPB5150A/DPB5700/DPB5700A
GSCH4000A

17