https://global.kikusui.co.jp/

P C R - L E / L E 2

AC Power Supply

S E R I E S

ofBroad range s
output mode

AC
DC
DC
AC+
High-performance multifunctional AC Power Supplies

PCR-LE/LE2 Series

Capable of various power line abnormality simulations and sequence operation.
Single–phase 500 VA to 9 kVA/Single–phase & three–phase 6 kVA, 9 kVA, 12 kVA, 18 kVA, 27 kVA,
Supporting the system for the single–phase, and expandable with optional drivers for the single–phase three–wire,
and three–phase operation.
Expandable capacity up to 27 kVA (single–phase), 54 kVA (single–phase three–line), and 81 kVA (three–phase)
Features a full range of measuring functions and supports AC, DC, and AC + DC Outputs
Detachable front panel
Eco-friendly function equipped
RS232C as a standard interface, GPIB, USB, and LAN () are available as an optional interface.

being smart

Solar
power

Wind
power

Fuel
cell

2

AC Power Supply

PCR-LE series

3New stage of AC power supply
supporting new energy field
High-performance AC Power Supplies

PCR-LE SERIESThe PCR-LE Series is a new line of advanced multifunctional AC power supply that has been developed

9 kVA

from our PCR-L/LA Series (linear amplifier type).
The PCR-LE Series provides high reliability andcan be applied to various applications, by taking advantage of the features that can control broadband
waveform freely. Moreover, the PCR-LE Series canbe configured as a core device of a test system combined with E-loads and Power Analyzers for “GridConnection Testing” in regard to dispersed power
generation, such as Solar Power, Wind Power, Fuel
Cell, and Gas Engine referred to as “New Energy
Field”. With various options, the low frequency immunity test and various power enviroment tests
are supported. The options for parallel operationand three-phase operation enable you to expand
a single-phase system up-to 27 kVA, single-phase
three wires up-to 54 kVA, and a three-phase system
up to 81 kVA. The system can be applied to a largescale EMC site for testing of industrial high-capacity
air conditioners.

▼[Applications]
Research & Development▼

Proof evaluation for power supply abnormality, EMC testing

Adjustment & Inspection Lines▼

Power supply voltage margin check, Automated inspection system

Production Lines▼

For stabilizing the line power supply, Automated testing system

Quality Assurance▼

IE ard Testing

After-Sales Service

As power supply for repair and calibration
To reproduce power line abnormalities
4

6 kVA

AC Power Supply

PCR-LE series● Lineup
Model

PCR500LE

PCR1000LE

Output capacity Single-phase 500 VA Single-phase 1 kVA
Maximum
output current
AC mode(L/H range)
DC mode(L/H range)

Dimensions(mm(inches))(Maximumdimensions)

Weight5 A / 2.5 A

10 A / 5 A

PCR2000LE

PCR3000LE

PCR4000LE

PCR6000LE

PCR9000LE

Single-phase 2 kVA

Single-phase 3 kVA

Single-phase 4 kVA

Single-phase 6 kVA

Single-phase 9 kVA

20 A / 10 A

30 A / 15 A

40 A / 20 A

60 A / 30 A

90 A / 45 A

60 A / 30 A

90 A / 45 A

1 V to 150 V / 2 V to 300 V
5 A / 2.5 A

10 A / 5 A

20 A / 10 A

30 A / 15 A

40 A / 20 A

±1.4 V to ±212 V / ±2.8 V to ±424 V
3.5 A / 1.75 A

7 A / 3.5 A

14 A / 7 A

21 A / 10.5 A

28 A / 14 A

42 A / 21 A

63 A / 31.5 A

430 (16.93") W

430 (16.93") W

430 (16.93") W

430 (16.93")(445 (17.52")) W

430 (16.93")(445 (17.52")) W

430 (16.93")(445 (17.52")) W

430 (16.93")(445 (17.52")) W

173 (6.81")(195 (7.68")) H

262 (10.31")(345 (13.58")) H

389 (15.31")(475 (18.70")) H

690 (27.17")(785 (30.91")) H

690 (27.17")(785 (30.91")) H

944 (36.17")1325 (52.17")(1040 (40.94")) H (1420 (55.91")) H

550 (21.65")(600 (23.62")) D

550 (21.65")(595 (23.43")) D

550 (21.65")(595 (23.43")) D

550 (21.65")(595 (23.43")) D

550 (21.65")(595 (23.43")) D

550 (21.65")(595 (23.43")) D

550 (21.65")(595 (23.43")) D

Approx. 17 kg(37.4 lbs)

Approx. 35 kg(77.1 lbs)

Approx. 55 kg(121.2 lbs)

Approx. 82 kg(180.7 lbs)

Approx. 96 kg(211.6 lbs)

Approx. 140 kg(308.6 lbs)

Approx. 190 kg(418.8 lbs)

Appearance

4 kVA
3 kVA

2 kVA
1 kVA
500 VA5

advantage

Production

Reliability
Inspection Production

R&D Service

Service

Production

Inspection
QA R&D
Service

R&D

R&D

Service

Inspection
Satisfy QA
SafetyService

The linear amplifier type realizes high stability and high quality output
and supports a wide range of functions from R&D to manufacturing/inspection lines and servicing.

What is a linear amplifier type?

What is a PWM inverter?

Firstly, the input power is converted to DC power by a rectifier circuit,then it supplies the power as the linear amplifier.
A sine wave reference voltage is created by such a crystal oscillator,
and it is used as input into the linear amplifier, where the power amplification is performed to generate the output power.
In addition to its high-speed response characteristics, because the output voltage and frequency can be changed whenever necessary, thissystem can be used to conduct simulations of power line abnormalities(such as instantaneous power interruption tests), and also it can be applied to the testing of ATE and other purposes.

This type uses a PWM (Pulse Width Modulation) switching-type
DC/AC inverter which is placed as a part instead of the linear
amplifier. Because this is a switching type, it cannot provide feedback over a wide range while the linear amplifier can. As a result,the output quality and response gets inferior, and noise becomeslarger, compared to the linear amplifier type.
However it has the advantages of being smaller and more efficientthan the linear amplifier type, and is also pulling attention as a highperformance AC power supply for energy-saving purposes.

List by PCR-LE applications
Mode

Category

Tested device
Home electronics,

Product tests
AC

AC + DCDC

AC,AC + DC,DC

6

office equipment,
industrial equipment

Test contents
Power fluctuation tests
IEC61000 standard low-frequency immunity tests

12 to 14

Reproduction and evaluation of voltage abnormalities in the market

Component
testsPower conditioners

Power regeneration tests

AC/DC converters

Power fluctuation tests

Component
tests

DC/DC converters

Tests of conversion from high voltage to low voltage
Simulations of voltage fluctuations in EV and HEV high-voltage batteries

Capacitors

Ripple current tests of high-voltage capacitors

EV charging systems

Tests of requirements for IEC61851 and ECE R10.04 standards

Component tests

Refer to page

12 to 13

14

AC Power Supply

PCR-LE series■ For R&D:

■ For Adjustment and Inspection lines:● Evaluation for the immunity of
power abnormalities.
Capableof DC output.
●● Easily conducting power measurement.
● Can be used in anechoic chambers and shield rooms.

● To confirm the power
voltage margin.
Use
in automated inspection systems.
●

The PCR-LE Series has equipped with the measurement functions built into the main unit, it can be used not only for voltage andcurrent measurement, but also for convenient measurement ofapparent and effective power, inrush (peak) current, power factor,
high-frequency current, and other values. Furthermore,it is capable
to conduct such as power line abnormality simulations, sequencing functions, and arbitrary waveform generation also provide a
dramatic improvement in data reproducibility and reliability whenevaluating immunity to instantaneous power interruptions, voltage
fluctuations, frequency fluctuations, missing phase, and other power line abnormalities. In addition, the PCR-LE has maximum DC
output of ±424 V. This is extremely convenient when a slight DC
output is required in case driving a DC/DC converter. The PCR-LE
Series can also be used as AC power sources in various EMC testsites (anechoic chambers, shield rooms, etc.).
* Use of the arbitrary waveform generation function and other functions requires
separate application software SD011-PCR-LE (Wavy for PCR-LE).

■ For Manufacturing lines:● Use as a CVCF power supply.
● Stabilization of the power line.
With the PCR-LE Series, it can be used as a CVCF power supply tohandle worldwide commercial power (100 V - 240 V), as well as formarine and aircraft power (400 Hz). It can supply a maximum output peak current up to 4 times the rating (rms) with a capacitor input
load (both peak value and continuous supply), or approximately 2times the rating (rms) for motors and other loads with large in-rushcurrents (peak value, approximately 10 seconds*, when power factor is 1). The PCR-LE Series is also recommended for power stabilization when using precision machining systems, measurement
systems, and others where the voltage abnormalities becomes anissue. With an output voltage response speed of 30 μs (standard
value) and a waveform distortion factor of 0.3 % or less, the PCRLE Series provides extremely high speed and high quality that are
particularly effective with systems such as welders and semiconductor manufacturing equipment where even slight power fluctuations or load fluctuations can affect quality and accuracy.

The PCR-LE Series can be used for operation checks of the power
voltage range, and as a power supply for aging. Multiple units of
the PCR-LE Series can be connected in parallel to boost capacity,and can also be connected in 3 phases, allowing flexible adaptation
to line changes or the number of aging units. Remote control andmonitoring from a PC is also supported using the GPIB or RS-232C
communication or USB or LAN interface, and it can be used for
management of inspection records and other quality data as well.
* The GPIB, USB, and LAN are available as an interface option.

■ For Quality Assurance:● Use as a standard room
power supply.
● Conducting of IEC standard tests.
The PCR-LE Series can be used as a power supply in standard
rooms and measurement device management rooms.

■ For After-sales service:● Use as a power supply
for repairs and calibration.
● Reproduction of power abnormalities.
The PCR-LE Series can also make a large contribution to repairs,
inspections, calibration, and other servicing work. For example, thePCR500LE (output capacity 500 VA) allows worldwide commercial
power (100 V - 240 V) to be supplied from a household electricaloutlet (100 V, 15 A). This is highly recommended for servicing sites
where large equipment cannot be installed and it also can be used
for the field service. Since the PCR-LE Series can supply clean
power that is free of fluctuation or distortion for inspection and calibration work, it can help to maintain and improve quality of service.

*Output shuts off after 10 seconds.

Waveform distortion occurs if the current exceeds the rating anytime during theperiod of 10 seconds.

7

features■ Extended system for large capacity applications. Flexible configuration in models.
■ Parallel operation *The separately-sold expansion operation driver is required.

It is possible to expand to 27 kVA (single phase), 54 kVA (single
phase 3-wire), and 81 kVA (three phase) by using the parallel,
single phase 3-wire, and three phase operation options (expansion
operation drivers). This allows the system to be used for large-scale
EMC site power or as test power for large-capacity industrial air
conditioners.

Can be expanded to 54 kVA (single phase 3-wire) or 81 kVA (three phase) when usedin combination with the single phase 3-wire option or three phase option.

★ Combinations of different models are possible!

Example: PCR2000LE + PCR4000LE + PCR6000LE = Single phase 12 kVA

12kVA

Standard
specification!

2kVA

PCR9000LE × 9 Rack

■ Single phase 3-wire, three phase operation* The separately-sold expansion operation driver is required.
All models / Max. expanded capacity: 54 kVA (single phase 3-wire),
81 kVA (three phase)■ Extensive configuration of the system.

When used in combination with the parallel operation option

Each unit can be used as either a master or slave, allowing units to
be individual or system depends on the requirement.

27
kVA
9kVA

9kVA

18
kVA
9kVA

9kVA

★ Combinations of different models are possible!

Example:PCR2000LE + PCR2000LE + PCR4000LE = 6kVA "Three-phase" or 8kVA
"Unbalanced Three-phase"

Singlephase

Single phase3-wire

Three
phase

+ 4kVA + 6kVA

Single
phase

27
kVA
9kVA

9kVA

9kVA

2kVA

* pThharesees6+ 2kVA + 4kVA

kVA

9kVA

*8kVA when used in the "Unbalanced Three-phase"■ For testing of the“Grid connected system” with reverse load flow
Conforming to the guideline of the Japanese standard requirements of system interconnection technologies

AC Power supply
PAT-T Series etc

Solar cell

Simulating
reverse load flow*

0-

V

Oscilloscope

DC Frequency
Powerconditioner

HarmonicAnalyzer KHA3000
HARMONIC /FLICKER ANALYZER

KHA3000

MEMORYMODEASSIST

HA

Vf

VIEW

START

STOP

UTILITY
OTHER

GPIB

HARDCOPYFILE

EXTCONTSYSTEM

REMOTELOCAL

KEYLOCK

F1
F2
USB

F3

AC Frequency

8

9

SHIFT

6

V RANGE I RANGE
1
2

3

ESC

F5

BUZZER
0

-

ENTER

F6
POWER

HOLD

ENTRY
7

V RANGE I RANGE
45F4REMOTEBUZZER.

BACKLIGHT

Line
impedance
network
LIN Series

LCR Load
Rotary plane load

AC Power supply PCR-LE Series

Load8

System
負荷*All the simulated reverse load flow power is consumed internally, thus, there will be no reverse load flow to the system.

AC Power Supply

PCR-LE series■ Wide-ranging specs
DC output also supported■ Eco-friendly function(Energy-saving function)■ Sleep function
The power unit goes into the sleep
mode when no output is detected
for a specified period to save thepower consumption.

■ Energy-saving operation function*
You can utilize the energy-saving
function to operate only the numberof power unit(s) depending on therequired supply load.
[Example] Operation with a 4 kVA model when 1 kVA
is necessary

PCR4000LE
PCR4000LE
PCR4000LE
PCR4000LE

SLEEP

SLEEP

OFF

OFF

OFF

OFF

OFF

OFF

ON

ON

Item

RatingVoltage (AC) *1

1 V to 150 V (L range), 2 V to 300 V(H range)

Frequency

1 Hz to 999.9 Hz *2

Voltage (DC/AC+DC) *1±1.4 V to ±212 V (L range), ±2.8 V to ±424 V (H range)*1: Settings available from 0 V.
*2: The frequency is limited to the range from 1 Hz to 500.0 Hz when the 3P05-PCR-LE(500Hz LMT) is installed in the PCR-LE series.

ln addition, the system supports a DC output mode and AC + DC
output mode. The system can be useful in a wider range of fields
such as chemistry- and physics-related areas.

■ Selectable response mode
Allows select of a response mode for the internal amplifier systemdepending on the load condition and application.
Item

Application

High-speed response (FAST)*3

for requesting a rate of power rise/fall

Normal response (MEDIUM) for testing various power supply environments
Highly stable response (SLOW) for power supply for EMC testing sites

Unit structure allows easy maintenance.

Maintenance (replacement or other work) on the power unit canbe performed in 1 KVA units. *Excepting PCR500LE■ Downsizing

PCR4000LA■ Power line abnormality simulation
In AC mode, it is possible to simulate power line abnormalities by setting
the output of the PCR-LE series system to the state of a power outage,
voltage drop (dip), or voltage increase (pop). This allows the ability totest switching power supplies and electronic equipment.

Comparison with the former model PCR-LA (4 kVA)
Existing model*3 : Excluding PCR6000LE, PCR9000LE, PCR6000LE2, PCR9000LE2, PCR12000LE2,
PCR18000LE2,PCR27000LE2, three phase operation, parallel operation

PCR4000LE

Size:

power outage

Downsized byapprox. 20 %Mass:
Decreased byModel

Dimensions (mm(inches))

Weight

PCR4000LE

445 (17.52") W×785 (30.91") H
×595 (23.43") D mm

96 kg (211.64 lbs)

PCR4000LA

455 (17.91") W×920 (36.22") H
×605 (23.82") D mm

120 kg (264.55 lbs)

■ Input/output terminal block tray
for easy connections
The rear input/output terminal block tray is a slide-out type,allowing input/output cables to be connected easily.

(Excepting the PCR500LE and PCR12000LE2 and PCR18000LE2 and PCR27000LE2)

When terminal block tray slides out*In case the terminal block tray is not returned into the storage compartment,
the PCR-LE2 can not be operated even if the power switch is turned on.

voltage drop (dip)■ External communication interface. Complied to LXI.

approx. 20 %Normal use

voltage increase (pop)RS232C (equipped as a standard). Remote control available with
GPIB, USB, and LAN as options. Using LAN makes it possible toconfigure highly cost-effective systems, as LXI standard is supported.

■ Other functions● Various measuring functions● Sequence function● Sensing● Regulation adjustment● Output current control● Setting output impedance● Measuring harmonics current● Soft start (Rise time control)● Internally fixed Vcc● Control panel angle adjustment

The control panel angle can be adjusted according to the position where it is used.
The optional control panel extension cable is also available. (See P. 18.)9

performance* Input power cord, load cable, terminal block, etc are also required for system build up.
Please make prior arrangements or consult your local Kikusui distributor. (Additional fee)

[Example of single phase 3-wire 4 kVA system]
Connecting cable (Accessories 75cm)● Example of single phase 3-wire system configuration					Capacity

Model

Qty

Single-phase three-wire driver

Qty

Single phase 3-wire 1 kVA

PCR500LE

2

2P05-PCR-LE

1

Single phase 3-wire 2 kVA

PCR1000LE

2

2P05-PCR-LE

1

Single phase 3-wire 4 kVA

PCR2000LE

2

2P05-PCR-LE

1

Single phase 3-wire 6 kVA

PCR3000LE

2

2P05-PCR-LE

1

Single phase 3-wire 8 kVA

PCR4000LE

2

2P05-PCR-LE

1

Single phase 3-wire 12 kVA

PCR6000LE

2

2P05-PCR-LE

1

Single phase 3-wire 18 kVA

PCR9000LE

2

2P05-PCR-LE

1

2P05-PCR-LE
V-phase board

2P05-PCR-LE
U-phase board

V

U

J4 J2

J1 J3

J4 J2

J1 J3

Power-sync cable (Accessories 1m)● Example of PCR2000LE parallel operation system configuration							Capacity

Model

Qty

Parallel operation driver (Master)

Qty

Parallel operation driver (Slave)Single phase 4 kVA

PCR2000LE

2

PD05M-PCR-LE

1

PD05S-PCR-LE

1

Single phase 6 kVA

PCR2000LE

3

PD05M-PCR-LE

1

PD05S-PCR-LE

2

Single phase 8 kVA

PCR2000LE

4

PD05M-PCR-LE

1

PD05S-PCR-LE

3

Single phase 10 kVA

PCR2000LE

5

PD05M-PCR-LE

1

PD05S-PCR-LE

4[Example of parallel operation 4 kVA system]

Qty[Example of parallel operation 6 kVA system]
Extension connection cable (PC01-PCR-LE, 1.3m)

Connecting cable (Accessories 75cm)

Connecting cable (Accessories 75cm)

PD05M-PCR-LE

PD05M-PCR-LE

PD05S-PCR-LE

Slave
J4 J2

J1 J3

J4 J2

PD05S-PCR-LE

MasterMaster
J4

J1 J3

J2

Power signal cable (Accessories 0.3m)

Connecting cable (Accessories 75cm)

Slave2

Slave1

J1 J3

J4

J2

J4

J1 J3

J2

Master
J1 J3

J4 J2

Slave1

J1 J3

J4 J2

J1 J3

Extension power signal cable (CC11-PCR-LE,1m)

Extension power signal cable (CC11-PCR-LE, 1m)● Example of PCR9000LE parallel operation system configuration

PD05S-PCR-LE

PD05M-PCR-LE

PD05S-PCR-LE

Extension power signal cable (CC11-PCR-LE, 1m)

			Capacity

Model

Qty

Parallel operation driver (Master)

Qty

Parallel operation driver (Slave)

Qty

Single phase 18 kVA

PCR9000LE

2

PD05M-PCR-LE

1

PD05S-PCR-LE

1

Single phase 27 kVA

PCR9000LE

3

PD05M-PCR-LE

1

PD05S-PCR-LE

2● Example of three-phase system configuration						Capacity

Model

Qty

Three-phase output driver

Qty

Three phase 1.5 kVA

PCR500LE

3

3P05-PCR-LE

1

Three phase 3 kVA

PCR1000LE

3

3P05-PCR-LE

1

Three phase 6 kVA

PCR2000LE

3

3P05-PCR-LE

1

Three phase 9 kVA

PCR3000LE

3

3P05-PCR-LE

1

Three phase 12 kVA

PCR4000LE

3

3P05-PCR-LE

1

Three phase 18 kVA

PCR6000LE

3

3P05-PCR-LE

1

Three phase 27 kVA

PCR9000LE

3

3P05-PCR-LE

1[Example of PCR2000LE Three phase 6 kVA system]
Connecting cable (Accessories 75cm)

3P05-PCR-LE
W-phase board

J4 J2

J1 J3

3P05-PCR-LE
W-phase board

3P05-PCR-LE
V-phase board

U
J4 J2

Power-sync cable (Accessories 1m)

W
J1 J3

J4 J2

* Illustration above is all rear panel.

3P05-PCR-LE
V-phase board

W
J1 J3

Power-sync cable (Accessories 1m)* The Master unit for power interlink in order to start up the equipment and the Master unit
controlling the system may differ when using the system configuration illustrated above.

10*1

Connecting cable (Accessories 75cm)

Connecting cable (Accessories 75cm)

3P05-PCR-LE
U-phase board

V

Extension cable (CC02-PCR-LE 2.8m)

J4 J2

3P05-PCR-LE
U-phase board

V
J1 J3

J4 J2

Power-sync cable (Accessories 1m)

U
J1 J3

J4 J2

J1 J3

Power-sync cable (Accessories 1m)*1: An optional extension cable (CC01-PCR-LE or CC02-PCR-LE) is available as needed according to
the unit layout.
* It is not possible to configure the system combined with the parallel operation and the three-phase
operation system. Please install the U-phase between the V-phase and the W-phase.

AC Power Supply

PCR-LE series● Example of parallel operation + Three-phase operation system configuration
Capacity

18 kVA

Model

Qty

Part

PCR2000LE9AC Power Supplies (2 kVA)

3P05-PCR-LE

1

Three-phase output driver

PD05M-PCR-LE

3

Parallel operation driver (Master)

PD05S-PCR-LE

6

Parallel operation driver (Slave)CC02-PCR-LE

2

Extension cable for 2P05·3P05 2.8 m

Capacity

81 kVA

Model

Qty

Part

PCR9000LE9AC Power Supplies (9kVA)

3P05-PCR-LE

1

Three-phase output driver

PD05M-PCR-LE

3

Parallel operation driver (Master)

PD05S-PCR-LE

6

Parallel operation driver (Slave)CC02-PCR-LE

2

Extension cable for 2P05·3P05 2.8 m[PCR2000LE 18 kVA example (Paralleled three-phase operation system)]
Extension connection cable(CC02-PCR-LE 2.8m)

Extension connection cable(CC02-PCR-LE 2.8m)

3P05-PCR-LE

➊

➊

PD05S

PD05S

Slave
J4 J2

Slave
J1 J3

J4 J2

Master
J1 J3

➋

PD05M

J4 J2

➊

➊

PD05S

PD05S

Slave
J1 J3

J4 J2

➌

➋

V-phase

Slave
J1 J3

J4 J2

PD05M

Master
J1 J3

➋

J4 J2

➊

PD05S

PD05S

Slave

J1 J3

J4 J2

➌

➋

➊

U-phase

Slave
J1 J3

J4 J2

PD05M

Master
J1 J3

➋

J4 J2

J1 J3

➋

W-phaseAccessories for three-phase driver and parallel operation driver➊Connecting cable (0.7m) ➋Power signal cable (0.3m) ➌Power-sync cable (Accessories 1m) *equivalent to the LC01-PCR-LE● Example of the combined system using different models
Capacity

15 kVA

Parallel operation system

Model

Qty

Part

PCR2000LE

1

AC Power Supplies (2 kVA)

PCR4000LE

1

AC Power Supplies (4 kVA)

PCR9000LE

1

AC Power Supplies (9 kVA)

PD05M-PCR-LE

1

Parallel operation driver (Master)

PD05S-PCR-LE

2

Parallel operation driver (Slave)PC01-PCR-LE

1

Extension connection cable (for parallel operation) 1.3 m

CC11-PCR-LE

2

Extension power signal cable (for parallel operation) 1 m

Capacity
1.5 kVA

Three phases expended system(11.5 kVA when using inthree-phase unbalanced)

Model

Qty

Part

PCR500LE

1AC Power Supplies (500 VA)

PCR2000LE

1

AC Power Supplies (2 kVA)

PCR9000LE

1

AC Power Supplies (9 kVA)

3P05-PCR-LE

1

Three-phase output driver

CC01-PCR-LE

2

Extension cable for 2P05·3P05 1.5 m[Example of 3 different-model units in parallel][Example of the three-phase unbalanced system]
3P05-PCR-LE

PD05M-PCR-LE

Extension connection cable (CC01-PCR-LE, 1.5m)

Connecting cable(PD05S Accessories 0.7m)

MasterPD05S-PCR-LE

Connecting cable (Accessories 0.75m)

PD05S-PCR-LE

Extension connection cable (PC01-PCR-LE, 1.3m)

Slave
Slave
J4 J2

Extension power signal cable (CC11-PCR-LE, 1m)
J1 J3

J1 J3

J4 J2

PCR2000LE

PCR9000LE

J4 J2

J1 J3
PCR4000LE

PCR9000LE

PCR2000LE

W

U

V

PCR500LE

Power-sync cable (Accessories 1m) *equivalent to the LC01-PCR-LE* Illustration above is all rear panel.

11

applications■ For testing of the Smart Grid related applications● Power conditioner (DUT)

Indoor

Applied to reverse
load flow!

Simulated power
supply for solar panels
PAT-TPower conditioner(DUT)

Load switch(route switch)

Power supply voltage
for detector circuit tests
PWR、PWX

SystemLine
impedance
networkAC

Fixed resistors

Power meter
KPM
Simulated AC load
PCZ

Simulated system
AC power supply
PCR-LE

Communication interface（LAN, USB, RS-232C, GPIB）● Storage Battery for Residential use (DUT)

IndoorStorage Battery for residential use（DUT）

Batteypower
conditioner

SystemSwitch BOX

Fixed resistorsLine
impedance
network

AC

Power meter
KPM
Simulated AC load

Simulated system
AC power supply
PCR-LE

PCZ
Communication interface（LAN, USB, RS-232C, GPIB）● EV charge and discharge station (DUT)EV

EV charge and discharge station(DUT)

IndoorLoad switch(route switch)

EV simulated
power supply
PCR-LE etc

DC

SystemLine
impedance
networkFixed resistors

AC

Power meter
KPM
Simulated AC load
PCZ
Communication interface（LAN, USB, RS-232C, GPIB）

12

Simulated system
AC power supply
PCR-LEAC Power Supply

PCR-LE series■ For testing of the Smart Grid related applications● Smart meter (DUT)

SystemIndoorRegenerative simulated
AC power supply
PCR-LE *Applied toreverse load flow!

Smart meter
（DUT）

Line
impedance
network

ResistBOX

AC

Simulated system
AC power supply
PCR-LE *Simulated load
PCZ

ACPower meter
KPM*Single-phase 3-wire driver
Requires the 2P05-PCR-LE

Communication interface（LAN, USB, RS-232C, GPIB）● Heat pump (DUT)

Heat pump unit
Compressor section

AC100VAC

AC or DC
Inverter

Motor

Compressor

Input
fluctuation test

Load variation
test

AC Power Supply

Electronic Load

DC Power Supply

PCR-LE

PLZ、PCZ

PAT-T

Drive test

Use as a simulated load for a power conditioner grid connection test system.

AC Electronic Load

PCZ1000A● Maximum input load power: 1000 W● Input voltage range: 14 V to 280 V(rms)● Input current range: 0 A to 10 A(rms)● Input frequency range: 45 Hz to 65 Hz● Equipped with constant current, constantresistance, and constant power modes.
● Parallel operation function(Max. 5 units, up to 5 kW/50 A rms)
● Equipped with tracking operation function● Crest factor function● RS232C equipped as a standard● Application software(option)

DUT
Solar panel

Power conditioner

AC powerTVAssimulated
load

Refrigerator

Air conditioner

Example: In-home alternatingcurrent loads

PCZ1000A

13

applications■ For Standard Compliance testing● Sihgle phase system● Three phase system

Three
phase

Singlephase

Harmonic/Flicker
AnalyzerLine
impedance
network

HARMONIC / FLICKER ANALYZER

KHA3000
ASSISTMEMORY

MODE
HA

Vf

UTILITY

GPIB

OTHER

HARD COPY

FILE

HOLD

EXT CONTSYSTEM

REMOTELOCAL

KEY LOCK
VIEW

START

F1
F2
USB

F3
F4

REMOTEF5

IEC dip
simulator

F6

STOP

ENTRY
7

8

9

SHIFT

V RANGE
4

I RANGE
5

V RANGE
1I RANGE
23ESC

BUZZER
0

.

-

ENTER

6

BUZZERBACKLIGHT

POWER

LINE IMPEDANCE NETWORK

LIN3060J
LINE IMPEDANCE NETWORK

ALARM

IMPEDANCE SELECT

ALARM

Harmonic/Flicker
Analyzer

LIN3060J

REMOTE

IMPEDANCE SELECTREMOTEAC Power supply
HARMONIC / FLICKER ANALYZER

KHA3000
ASSISTMEMORY

MODE
HA

Vf

UTILITY

PCR-LE

GPIB

OTHER

HARD COPY

FILE

HOLD

EXT CONTSYSTEM

REMOTELOCAL

KEY LOCK
VIEW

START

F1
F2
USB

F3
F4

REMOTE

F5
F6

STOP

ENTRY
7

8

9

SHIFT

V RANGE
4

I RANGE
5

V RANGE
1I RANGE
23ESC

BUZZER
0

.

-

ENTER

IEC dip
simulator

6

BUZZERBACKLIGHT

POWER

AC Power supply

PCR-LE

POWER

Line
impedance
network

This system can simulate various conditions of phenomena occurringin AC power environments. It can be used for immunity tests ofelectrical and electronic devices which are connected to a low-voltage
distribution system, or which have DC power input ports, under the
standard conditions as specified on the right. The test conditions canbe set outside the standard range, allowing the system to be used forpreliminary tests prior to standard tests, immunity margin tests, andstress tests. The KHA3000 harmonic/flicker analyzer combines a PCRLE Series AC power supply, LIN Series line impedance network, andapplication software*, allowing tests which conform to IEC standards
and JIS standards.
*SD009-PCR-LE/WE [Quick Immunity Sequencer 2] is required. (See P. 16.)

POWER

● IEC61000-4-11...................Voltage dipping, instantaneous power
failure and voltage variation● IEC61000-4-13...................Higher harmonics wave/interharmonic wave
● IEC61000-4-14...................Voltage swing● IEC61000-4-27..................Unbalance in units● IEC61000-4-28..................Variation in power supply frequency
for units with 16 A/phase● IEC61000-4-34..................Voltage drop (dip), instantaneous power
failure and voltage variation for units
with input current exceeding 16 A/phase*● IEC61000-4-17...................Ripple at the DC input power terminal● IEC61000-4-29..................Voltage drop (dip), instantaneous power
failure and voltage variation in DC*● IEC61000-3-2,12................Harmonic electric current limit level● IEC61000-3-3,11................Voltage fluctuation,Flicka limit level* Designed for preliminary test purposes. For details, please refer to page 15 and 16.

■ For testing of the EV charging system
Power suppy line interference
Quick lmmunity sequencer
SD009-PCR-LE/WE● EV charging system (Item under test)
Quick charge

Residential charge

Electronic load
PLZ-4WH Series

High-performance AC Power Supply
PCR-LE SeriesHigh voltage
battery system

DC
DC400V

High-performance
AC Power Supply
PCR-LE Series

14

AC100V/200V

controllerAC

EV charging
systemLow voltage
battery

IEC dip
simulator

AC Power Supply

PCR-LE series

IEC Dip·Simulator

DSI Series [DSI1020/DSI3020]
For the Voltage dips, short interruptions and voltage variations
immunity test system, complied to the IEC61000-4-11 (2004)
The DSI Series is an option unit used to configure the test system complying with the “Voltage Dips, Short
Interruptions and Voltage Variations Immunity Tests” as defined in the IEC61000-4-11 (2004)
standard. It can be used in combination with the Kikusui AC power supplies (PCR-LE/LE2 series). It meetsthe test requirement of : high-speed voltage switching (rise time: 1 μs to 5 μs), voltage dips (0 %, 40 %, 70 %,and 80 %), and phase-voltage and line-voltage tests.

▼▼▼▼■ DSI1020 : Applied to the Single-phase two-wire system■ DSI3020 : Applied to the Single-phase two-wire, Single-phase three-wire,
Three-phase three-wire, and Three-phase four-wire system.
Fast Votage rise/fall time (1 µs to 5 µs)
Applied to the voltage dips (0 %, 40 %, 70 %, and 80 %)Applied to the Line Voltage-dip* and the Phase Voltage-dip
Maximum Line Input voltage 500 V (rms)▲* The Line Voltage-dip applied to only the "DSI3020".

When connecting the DSI Series with the PCR-LE Series, the output capacity of the AC power
supply of each phase will be limited. For details, please refer to the individual product brochure orcontact our local distributor.

DSI3020

Wiring configuration

Model

Maximum current(per phase)

Single phase

DSI1020

16 A✔

DSI3020

16 A✔

DIP level

Complied standard

Remarks

0/40/70/80 %IEC61000-4-11 (2004)

For Single Phase only

0/40/70/80 %IEC61000-4-11 (2004)

For Single Phase or Three Phase

Three phase✔

Line Impedance Network

LIN Series [LIN1020JF/LIN3020JF/LIN3060J/OP01-LIN1020JF]
It is equipped with the IEC/JIS standard impedance. It supports voltage fluctuation and flicker tests.
■ LIN1020JF

LIN1020JF is equipped with the impedance determined by the IEC flicker test (IEC61000-3-3) and JIS harmonics(JIS C61000-3-2), which can be configured via the USB interface (standard feature) or the contact signal interface
from the application software. The single-phase two-wire IEC flicker/harmonics test system can be configured in
combination with AC power supply PCR-LE/LE2 and harmonic flicker analyzer KHA1000/KHA3000.

■ LIN3020JF
LIN3020JF is equipped with the impedance determined by the IEC flicker test (IEC61000-3-3) and JIS
harmonics (JIS C61000-3-2), which can be configured via the USB interface (standard feature) or thecontact signal interface from the application software. The single-phase two-wire/three-wire/three-phase
IEC flicker/harmonics test systems can be configured in combination with AC power supply PCR-LE/LE2
and harmonics flicker analyzer KHA1000/KHA3000.

■ OP01-LIN1020JF▲

OP01-LIN1020JF is an additional unit that is used to expand LIN1020JF in three phases (addition of V phase and W phase).
LIN3020JF■ LIN3060J
The LIN3060J is an essential reference impedance unit for building grid-connected power conditioner test
systems.
* Note that this is not applicable to the IEC flicker test. Contact us for a product that is compliant with IEC61000-3-11.

Model

Maximum
current(per phase)

Complied standard
Wiring configuration

IEC 61000-3-3
230 V 50 Hz

Remarks

JIS C61000-3-2 *1
100 V 50/60 Hz

200 V 50/60 Hz

Single phase 2-wire

✔

✔

✔

Product for IEC flicker /voltage fluctuation test

Single phase 2-wire/3-wireThree phase 3-wire/4-wire

✔

✔

✔

LIN1020JF+
OP01-LIN1020JF *2

Single phase 2-wire/3-wireThree phase 3-wire/4-wire*1 Insertion of the impedance is optionalin the JIS harmonics test. (Normallyapplied for bypass.)

✔

✔

✔

*2 OP01-LIN1020JF does not work solely.

LIN3060J

Single phase 2-wire/3-wireThree phase 3-wire/4-wire✔

✔

Product for grid connection test

0.4 Ω + 0.37 mH(Z1)0.38 Ω + 0.46 mH(Z2)

LIN1020JF
LIN3020JF

Impedance Value

20 A

60 A

Single phase 2-wireSingle phase 3-wireThree phase 3-wireThree phase 4-wire

0.4 Ω + Jn0.25 Ω(Z3)0.19 Ω + 0.23 mH
0.19 Ω+0.23 mH
0.24 Ω + Jn0.15 Ω(0.16 Ω+Jn0.1 Ω for N phase) (0.21 Ω + Jn0.14 mH for N phase) (0.19 Ω + Jn0.23 mH for N phase)

15

options■ Application software

EMC

IEC61000-4[Caution] For customers using the former PCR-L/LA Series
Please be aware that the PCR-LE Series is not interchangeable with the former PCR-L/LA Series of products.
Therefore it is not possible to upgrade a system with a combination of products from the two different series'. In
general (with some exceptions) the options from one series cannot be used in the other. If there are any unclear
points or for other details, please contact a Kikusui sales office.

* For details, please see the Kikusui homepage.

Power Line Disturbance Immunity Testing Software

SD009-PCR-LE/WE [ Quick Immunity Sequencer 2 ]List of conformance to the EMCstandard tests✔ : Conforming as standard▲ : Partially non-conforming

The latest standards for IEC61000-4 supported!

– : Function not available

Standard

Item

IEC61000-4-11

Voltage dip

Conforming
Single-phase

Three-phase

✔ *1
✔ *1
✔
✔
✔
✔
✔
✔
✔
✔
✔
✔
✔
–
–

IEC61000-4-14

Voltage fluctuation

Voltage fluctuation

Interval

IEC61000-4-17

Single-phase rectifier circuit

Ripple on d.c. input power port

Three-phase rectifier circuit

✔ *1
✔ *1
✔
✔
✔
✔
✔
✔
✔
✔
✔
✔
✔
✔
✔

IEC61000-4-27 Unbalance
IEC61000-4-28

Unbalance–▲*2

Frequency variations✔

✔

IEC61000-4-29

Voltage dips

✔
▲*3
✔
▲*4
▲*4
✔

–
–
–
▲*4
▲*4
✔

Voltage dips, short interruptions, and voltage
variations

Short interruption
Voltage variation
Flat curve
Over swing
Sweep in frequency

IEC61000-4-13

Odd, non multiple of 3

Harmonics and interharmonics

Odd, multiple of 3
Even harmonics
Interharmonics
Meister curve

Variation of power frequency

Voltage dips, short interruptions, and voltage
variations on d.c. input power port

Short interruptions

IEC61000-4-34

Voltage dips

Voltage dips, short interruptions, and voltage
variations

Voltage variations
Short interruptions
Voltage variations“Quick Immunity Sequencer 2” (model name: SD009-PCR-LE/WE) is an application
software for immunity testing with the AC power supply PCR-LE series system,
based on the power line disturbance standard (IEC61000-4 Series) for the
immunity testing of the EMC standard.
Not only can it be used for compliance testing based on the latest standards orfor some types of preliminary testing, but the software can be also employed
for advance checking in development phases and for immunity margin tests,
because it allows extended testing conditions to be set as needed.

* Immunity testing for units with 16 A/phase except for those required by IEC61000-4-34*1 Conforms to the standard when used in combination with DIP Simulator. If using the PCR-LE/LE2 alone, the voltage dipsand short interruptions are preliminary tests.
*2 Capability of rapid change with 1 µs to 5 µs is required for 110 %, 95.2 %, 93.5 %, 90 %, 87 %, 80 %, 74 %, 71 %, 66 %.
Preliminary test is capable since the voltage response of the PCR-LE/LE2 is 20 µs in FAST mode and 30 µs in MEDIUM mode.
*3 Must support output impedance greater than 100 kΩ. The PCR-LE/LE2 output impedance is less than 100 kΩ and therefore designed for preliminary testing purposes.
*4 The device between the range of 16 A to 75 A requires to have the capability of rapid change with 1 μs to 5 μs.
The device exceeding 75 A does not require to have the capability of rapid change with 1 μs to 5 μs. (It is relaxed to 1 µs to 50 µs for the device exceeding 75 A.)

Remote Control Software for the Windows Tablet

SD021-PCR-LE/WE [ RMT CONT SOFTWARE FOR PCR-LE/WE ]The Windows tablet can be used as a remote controller !
The SD021-PCR-LE/WE is the software that can control the PCR-LE/LE2 Series. It iscapable to change the setting condition of the "wiring method", "output mode",

Screen lockCommunication"voltage range", "voltage value", and "frequency value". And these settings changedby the remote controller can be saved and recalled. Moreover, it can display themeasurement value of the AC power supply. The remote operation and control of
the AC power supply from the distance can be easily realized.
● Operating Environment : Intel Core i5 or better / Windows 10 or Windows 8.1 / Memory 4GB /10 GB or more free hard disk space / Display resolution 1366×768 dots or better / USB port
*The LAN cable, LAN adaptor (micro USB to the wired LAN), the optional LAN board (LN05-PCRLE) are required.

Voltage value

Wiring method

Frequency value

Output mode
Voltage range

OutputMonitor

Screen display (main screen)16

AC Power Supply

PCR-LE series■ Application software

Wavy

Trial version
is available
on our web!!

“Wavy” Sequence Creation Software

https://global.kikusui.co.jp/
downloads/

SD011-PCR-LE [ Wavy for PCR-LE ]Download !

The software extends the feature of waveform generation and sequencefunctions. Easy sequence control without programming knowledge.
Wavy is an application software that supports sequencecreation and the operation for Kikusui power supplies andelectronic loads.
Wavy allows you to create and edit sequences visuallywith a mouse without programming knowledge. Real-▲Commandcontrol

time monitor function is added to the Ver. 4.0 or later, thatenables monitoring and logging values of voltage andcurrent. The Ver.5.0 equips Remote Control Panel function▲Direct control▲

Graph viewer/Configurationthat enables you to control power supplies as if you wereusing a remote controller.

■ It makes you easier to create or edit the test condition file required for the sequence operation.
■ By using the storage function of test condition data file,
it enables you to manage the test condition of the standard routine test.
■ The progress of execution sequence will be displayed on the "practical dialogue" with the setting value and the cursor.
■ It is possible to observe the intuitionistic output through by the "monitor graph" that plots the ongoing monitor value.
■ You can save the acquired monitor data as a test result.
■ Added the "waveform image" window. You can easily keep track of the AC signal.
■ Allows you to edit and create the new arbitrary waveform easily. You can instantly write then output the created arbitrary waveform.
■ Supports the status of description of sequence step for "selected" or "not selected". It enables you to select depends on the requirement such as the"pausing function", "trigger function", or "AC waveform".
■ Newly added features of "Sequence Pre-view Dialog" enables you to confirm the waveform before executing the sequence operation.

Avionics Test Software

SD012-PCR-LE/WE

Supported Standards

Supporting to the compliance testing of the avionics test standard.
The test pattern can be conducted from the Library.
PCR-LE Output SettingStandard Selection
Step List Editing Screen

Step List

Test Selection

Test Run

Output Monitor
StatusClick Button
to Run Step

[Main Screen][Test Details (Step List) Editing Screen]■ Easy configuration - just select standard from library■ Test step editing and saving - convenient for development and evaluation required with marginal testing■ Test condition reporting function - enables test history logging■ Remote control via LAN

Military Standard:MIL-STD-704A/E/F
Civilian Standard:RTCA DO-160F/G
Civilian Standard:JIS W0812:2004

Test standards have been established that electricalcomponents and parts installed on aircraft must meet. All
electrical components and parts installed on the fuselage
must comply with these standards, but the applicable
test standards vary according to the intended use andpurpose. Test standards can be largely divided into two
types: military standards and civilian standards. In addition,
aircraft manufacturers sometimes apply their own set ofprivate standards.Avionics Test Software [SD012-PCR-LE/
WE] is a software application that support to the aircraft
test standards, and is used to control the PCR-LE/LE2 Series
that enables you to conduct the test standards for the
MIL-STD-704, RTCA/D0-160 and JIS W0812 standards. Test
patterns are library-based, which enables tests to be easily
run by simply selecting the wiring configuration and thetype of test.In general, the 400 Hz AC power supply is used
for the large aircraft, and the 28 V DC power supply is used
for the small aircraft

17

options■ Interface boards

GPIB

USB

* LE2 indicates the available option for the multi-output models, "PCR-LE2 Series".

* Any one of the following can be installed.

GPIB Interface LE2
IB05-PCR-LELAN

USB Interface LE2
US05-PCR-LE
LAN Interface (LXI) LE2
LN05-PCR-LE* Any one of the following can be installed.

■ Analog signal interface boards
EX05-PCR-LE* (An Amplifier type) LE2

EX06-PCR-LE (Amplitude control type)

Amplifies the input waveform without changing it. By using
this interface board, you can control the PCR-LE with an
external contact for (output ON/OFF, sequence start/stop,
alarm clear, forced power OFF) and operation status monitoring (output status, alarm status, busy status, current
peak limit and overload status).

The output AC voltage value can be varied according to
the input voltage signal. By using this interface board,
you can control the PCR-LE with an external contact
for (output ON/OFF, sequence start/stop, alarm clear,forced power OFF) and operation status monitoring(output status, alarm status, busy status, current peak
limit and overload status).

Note: If the input waveform will be amplified and used in a multiphase system, one of these interface board is required for eachphase.PCR6000LE2 and PCR9000LE2 cannot amplify the input
waveform in multi-phase output mode.

EX05-PCR-LEModel

The input waveform is directly
amplified and output.

EX05-PCR-LE

Example: 1 Vrms
AC Sine wave

AmplifierOutput100 Vrms AC Sine waveor
200 Vrms AC Sine wave

Voltage amplification rate: 100-times
or 200-times

EX06-PCR-LE

Output Wirings

Required Quantity

PCR-LE Series

PCR-LE2 Series

Single-phase two-wire

1

PCR-LE Series

PCR-LE2 Series

Single-phase three-wire2

U-phase,V-phase

U-phase,V-phase *Single-phase three-wire/four-wire3

U-phase,V-phase,W-phaseU-phase,V-phase,W-phase *Single-phase two-wire

1

PCR-LE Series

PCR-LE2 Series

Single-phase three-wire

1

Single-phase three-wire/four-wire

1

EXT-AC mode

The voltage of the output alternating
current can be changed based on
the level input DC signal.

Input
Sine wave 10 V

U-phase0 V to 135 V ACor
0 V to 270 V AC

U-phaseVoltage amplification rate: 13.5-times
or 27-times* LE2 indicates the available option for the multi-output models, "PCR-LE2 Series".

For PCR1000LE
3-core cabtire cables 5.5 mm2 /3 m M4
AC5.5-3P3M-M4C

For PCR4000LE
3 single-core cables 22 mm2 /3 m M8
AC22-1P3M-M8C-3S

For PCR2000LE
3 single-core cables 8 mm2 /3 m M5
AC8-1P3M-M5C-3S

For PCR9000LE/PCR9000LE2 LE2
4 single-core cables 14 mm2 /3 m M5
AC14-1P3M-M5C-4S

For PCR3000LE/PCR6000LE/PCR6000LE2 LE2
3 single-core cables 14 mm2 /3 m M8
AC14-1P3M-M8C-3S

Power-sync cable,1 m
Multiple units of the PCR-LE Series can be connected and turned ON/OFF.
LC01-PCR-LE■ Control panel cable
Extension cable for control panel
EC05-PCR (cable's length: 2 m)

* LE2 indicates the available option for the multi-output models, "PCR-LE2 Series".

EC05-PCR

LE2
Desk
PCR2000LE

Neatly stored
under the desk
475 mmUse image*Depth: 595 mm

18

OutputAmplifierExample: Input DC
voltage0 to 10 V*The PCR6000LE2 and PCR9000LE2 do not have a feature to amplitude the input waveform in the multiple output mode.

■ Input power cord/Power-sync cable

LE2

EX06-PCR-LE

				

EXT-DC mode

Input

* LE2 indicates the available option for the multi-output models, "PCR-LE2 Series".

Image of using EC05-PCR

430 mmAC Power Supply

PCR-LE series■ Parallel operation driver
PD05M-PCR-LE

Note: When using this product, a PCR-LE Series unit with firmware version 3.01 or later is required.
If the firmware of your product is 1.X or earlier, modifications and other changes will be required. Please consult with yourlocal distributor. This option cannot be used with PCR500LE or PCR1000LE.

Parallel operation driver (Master)
PD05M-PCR-LE
Parallel operation driver (Slave)
PD05S-PCR-LE
Accessories: Connecting cable (0.7 m), Power signal cable (0.3 m)
PD05S-PCR-LE

Extension cableThis extension cable is used if the provided connection cable (0.7 m) or power signal cable is too short whenthe master unit layout is changed or when connecting different models together.

Extension connection cable (1.3 m) PC01-PCR-LE*Used between models with a power difference of 4 kVA or more.

Extension power signal cable (1 m) CC11-PCR-LE*Used when the placement of Master and Slave devices are reversed.

■ Single-phase 3-wire output /Three-phase output driver* A single-phase 3-wire output driver and three-phase operation output driver cannot be used in combination.
2P05-PCR-LE

Note: When using this product, the PCR-LE Series unit with firmware version 2.0 or later is required.
If the firmware of your product is 1.X or earlier, modifications and other changes will be required. Please consult with your local distributor.

Single-phase 3-wire output driver
2P05-PCR-LE
Accessories : Connecting cable (0.75m), Power-sync cable (LC01-PCR-LE, 1 m)

Three-phase output driver/Three-phase output driver (500 Hz limit type)3P05-PCR-LE/3P05-PCR-LE (500Hz LMT)
Accessories : Connecting cable (0.75 m)×2, Power-sync cable (LC01-PCR-LE, 1 m) ×2
3P05-PCR-LE

Extension cableThis extension cable is used if the provided connection cable (0.75 m) is too short when connecting different
models together or when using the parallel operation driver.

Extension connection cable (1.5 m) CC01-PCR-LE*Used between models with a power difference of 2 kVA or more, as well as in cases where two units are paralleled
per phase for paralleled three-phase operation. CC02-PCR-LE is required in cases where a model smaller than the
PCR2000LE is used for a three-phase operation system with PCR9000LE.

Extension connection cable (2.8 m) CC02-PCR-LE*Used for paralleled three-phase operation system where three units or more are in parallel per phase.

■ Rack mount/Prodout about standard
For PCR500LE Brakets

Base holding angle

For PCR1000LE Brakets

SPEC40414A

KRB4 (For EIA inch size)
KRB200 (For JIS metric size)

KRB6 (For EIA inch size)
KRB300 (For JIS metric size)

For PCR2000LE Brakets
KRB9 (For EIA inch size)
KRB400 (For JIS metric size)OP03-KRC

Residual charge measurement

This unit is applied to the residual charge measurement in conformance with theElectric Appliance Safety Law, IEC60950-1, IEC60335-1, IEC60065, and otherregulations. It allows residual charge to be measured easily and accurately without
unplugging work.

19

exterior design■ Front panel
PCR500LE
RS232C CONNECTORCONTROL PANELUSB CONNECTOR

Adjustable angle, removable* Can be used as a remote controller
when the optional extension cable isused. (See P. 18.)

USB memory connector* Not used for control.

POWER SWITCH
Power ON/OFF switch

PCR2000LE

POWER UNITS
AND AIR INLETOUTPUT

Contains the intake port and dust
filter for internal cooling of thepower unit.
* The eco-friendly function(energy-saving operation function)
is not applied with PCR500LE or
PCR1000LE.

RS232C CONNECTORCONTROL PANEL
Adjustable angle, removable* Can be used as a remote controller
when the optional extension cable isused. (See P. 18.)

USB CONNECTOR
USB memory connector* Not used for control.

POWER UNITS
AND AIR INLET

Contains the intake port and dust
filter for internal cooling of thepower unit.
* The eco-friendly function(energy-saving operation function)
is applied with PCR2000LE(2 kVA) or more.(See P. 9.)

CIRCUIT BREAKER
Outlet breaker reset button

OUTPUT

20

POWER SWITCHPOWER SELECTOR SWITCH

Power ON/OFF switch

This switch is used to select between the master unit
and the slave unit operation when multiple PCR-LE
units are used.

AC Power Supply

PCR-LE series■ Rear panel
PCR500LESEQ STAT OUT CONNECTORSEQ TRIG IN CONNECTOR

EXPANSION SLOTFor a trigger input(BNC)

For an analog signal interface

For a status output(BNC)EXPANSION SLOTSEQ TRIG OUT CONNECTOR

EXPANSION SLOTFor a master-slave parallel operation (Not available for PCR500LE.)

For a multipul-phase operation

For a trigger output(BNC)EXPANSION SLOTFor a communication interface

AC INPUTSENSING TERMINAL BLOCKVENT HOLE

Connect the sensing cable

POWER SELECTOR SWITCHOUTPUT TERMINAL BLOCK(With cover)

EXPANSION SLOTFor a multipul-phase operation

EXPANSION SLOTFor an analog signal interface

EXPANSION SLOTFor a master-slave parallel operation(Not avalable for PCR500LE.)

PCR1000LE
EXPANSION SLOTFor a communication interface

PCR9000LE
Input/output terminal block tray
(Excluding the PCR500LE)OUTPUT TERMINAL BLOCKINPUT TERMINAL BLOCKSENSING TERMINAL BLOCK

Slide-out structure allows wiring easily.

Connect the sensing cable

PCR6000LE

PCR3000LE

PCR4000LE

Normal use

PCR2000LE

When the terminal block tray slides out
Photo : PCR1000LE*In case the terminal block tray is not returned to theposition in the storage compartment, the PCR-LE can
not be operated even if the power switch is turned on.

21dimensions

430 (16.93)

0°12340 (13.39)

0°

MAX 20(0.79)

48 (1.89)

457 (17.99)

12

430 (16.93)

51.2 (2.02)

12 M6 screw holes,
max. depth 10 mm (0.39 ”)

550 (21.65)
MAX 600 (23.62)

550 (21.65)

MAX 20(0.79)

A-arrow caster screw position diagram

MAX 595 (23.43)

23.2(0.91)

262 (10.31)

390 (15.35)

37(1.46)

385 (15.16)

84
50
50
(1.97) (3.31) (1.97)

MAX 345 (13.58)

115 (4.53)

173 (6.81)

8 M4 screw holes, max. depth 10 mm (0.39 ”)

4 M4 screw holes,
max. depth 10 mm (0.39 ”)

27.5(1.08)

MAX 195 (7.68)

4 M4 screw holes, max. depth 10 mm (0.39 ”)

85 (3.35)

A arrow

26 (1.02)

PCR1000LE

550 (21.65)

MAX 20(0.79)

A-arrow caster screw position diagram

16 M6 screw holes,
max. depth 10 mm(0.39 ”)

342 (13.46)

0°

12 M6 screw holes,
max. depth 10 mm (0.39 ”)

430 (16.93)

0°12340 (13.39)

50(1.97)

48 (1.89)

61.2(2.41)

430 (16.93)

457 (17.99)

1251.2 (2.02)

440 (17.32)

50(1.97)

59.2(2.33)

MAX 445 (17.52)

PCR500LE

550 (21.65)
MAX 595 (23.43)

MAX 20(0.79)

A-arrow caster screw position diagram

MAX 595 (23.43)

26 (1.02)

A arrow22(0.87)

45.5(1.79)

100(3.94)

76.5(3.01) 195(7.68)

176.5(6.95)

MAX 785 (30.91)

389 (15.31)
100
100(3.94)(3.94)

MAX 475 (18.70)

8 M4 screw holes, max. depth 10 mm (0.39 ”)

690 (27.17)

195 (7.68)

16 M4 screw holes, max. depth 10 mm (0.39 ”)

M12 screw hole, max. depth 35 mm (1.38 ”)
26(1.02)

PCR2000LE

22

PCR3000LE

PCR4000LE

A arrow
140.2 40
(5.52) (1.57)

200 (7.87)
508.5 (20.02)

40(1.57)
8 M10 screw holes,
max. depth
10 mm (0.39 ”)

AC Power Supply

PCR-LE series
16 M6 screw holes,
max. depth 10 mm (0.39 ”)

MAX 445 (17.52)
430 (16.93)

342 (13.46)

50

50(1.97)(1.97)

440 (17.32)

59.2(2.33)

61.2(2.41)

A-arrow caster screw position diagram

550 (21.65)
MAX 595 (23.43)

MAX 20(0.79)

257.5(10.14)
76.5
22 (3.01)(0.87)

250 (9.84)

MAX1040 (40.94)944 (36.17)

250 (9.84)

16 M4 screw holes, max. depth 10 mm (0.39 ”)

M12 screw hole, max. depth 35 mm (1.38 ”)

PCR6000LE

440 (17.32)(1.97)
50

200 (7.87)

40(1.57)

8 M10 screw holes,
max. depth
10 mm (0.39 ”)

508.5 (20.02)

16 M6 screw holes,
max. depth 10 mm(0.39 ”)

61.2
(2.41)

A-arrow caster screw position diagram

430 (16.93)

342 (13.46)

MAX 445 (17.52)

50(1.97)

59.2(2.33)

A arrow

140.2 40
(5.52) (1.57)
26 (1.02)

MAX 20(0.79)

550 (21.65)
MAX 595 (23.43)

195 (7.68)
195 (7.68)
22(0.87)

76.5(3.01)

195 (7.68)

195 (7.68)

MAX 1420 (55.91)
1325 (52.17)

195 (7.68)

195 (7.68)

28 M4 screw holes, max. depth 10 mm (0.39 ”)

PCR9000LE
M12 screw hole,max. depth 35 mm (1.38 ”)

26(1.02)

A arrow
140.2(5.52)

40(1.57)

200 (7.87)

40(1.57)
508.5 (20.02)

8 M10 screw holes,
max. depth
10 mm (0.39 ”)

23specifications
Item/Model

PCR500LE

PCR1000LE

PCR2000LE

Input ratings (AC rms)

PCR3000LE

PCR4000LE

PCR6000LE

1P2W

3P3W200V

Voltage

85 V to 132 V /170 V to 250 V *1

PhasesSingle phase

170 V to 250 V

PCR9000LE
3P4W400V

3P3W200V

3P4W400V

Line voltage
324 V to 440 V(Phase voltage 187 V to 254 V)

170 V to 250 V

Line voltage
324 V to 440 V(Phase voltage 187 V to 254 V)

Three phase 3-wires Three phase 4-wires Three phase 3-wires Three phase 4-wires

Frequency

47Hz to 63Hz

Apparent power

Approx. 0.93 kVA Approx. 1.8 kVA Approx. 3.6 kVA Approx. 5.5 kVA Approx. 7.3 kVA

Power factor *2
Max. current *1
AC mode output ratings (AC rms)

11.3 A, 5.5 A

22 A, 10.8 A

44 A, 21.5 A

66 A, 32 A

Voltage (output L range, output H range)

38 A

Approx. 15.7 kVA
21 A

55 A

30 A

1 V to 150 V / 2 V to 300 V

Resolution
Voltage setting range
Voltage setting accuracy (output
L range, output H range) *3
Max. current (output L range,
output H range) *4

Approx. 10.6 kVA

0.97 (TYP)
88 A, 43 A
64 A

0.1V
0 V to 152.5 V / 0 V to 305.0 V± (0.3 % of set + 0.6 V)
5 A, 2.5 A

10 A, 5 A

20 A, 10 A

30 A, 15 A

40 A, 20 A

500 VA

1 kVA

2 kVA

3 kVA

4 kVA

Phase

60 A, 30 A

90 A, 45 A

6 kVA

9 kVA

Single phase

Power capacity
Maximum peak current *5

Max. current (rms) × 4 (TYP)

Max. reverse current *630 % of the max. current (rms)

Load power factor

0 to 1 (leading or lagging) *4

Frequency *4

1 Hz to 999.9 Hz
Resolution

0.01 Hz (1.00 Hz to 100.0 Hz), 0.1 Hz (100.0 Hz to 999.9 Hz)

DC mode output ratings
Voltage±1.4 V to ±212 V / ±2.8 V to ±424 V
Resolution

0.1 V

Voltage setting range-215.0 V to +215.5 V / -431.0 V to +431.0 V

Voltage setting accuracy (output L range,
output H range) *7
Max. current *8±(0.05 % of set + 0.05/0.1 V)
3.5 A, 1.75 A

7 A, 3.5 A

14 A, 7 A

21 A, 10.5 A

Max. instantaneous current *9
Power capacity

28 A, 14 A

42 A, 21 A

63 A, 31.5 A

4.2 kW

6.3 kW

Max. current (rms) × 3.6
350 W

700 W

1.4 kW

2.1 kW

2.8 kW

Output voltage stability
Line regulation *10

Within ±0.1 %Load regulation (output L range,
output H range) *11
Output frequency
variation *12

Within ±0.1 V, within ±0.2 V

FASTWithin ±0.2 %–MEDIUM

Within ±0.3 %Ripple noise in DC mode (5 Hz to
1 MHz components)
Ambient temperature variation*13

0.15 Vrms or less

0.2 Vrms or less0.25 Vrms or less
100 ppm/°C (TYP)

Output frequency stability, output voltage waveform distortion ratio, output voltage response speed, efficiency
Within ±5×10- 5

Output frequency stability *14
Setting
accuracy

Within ±1×10- 4

FAST
Output voltage waveform
distortion ratio *15
MEDIUM
Output voltage
response speed *16–±0.2 % or less±0.3 % or less

FAST

–

20 μs (TYP)

MEDIUM

30 μs (TYP)
54 % or more,
56 % or more

Efficiency *17

55 % or more, 57 % or more

58 % or more

Meters (fluorescent display)
Voltmeter *18
Ammeter *18
Wattmeter *19
*1
*2
*3
*4

24

Resolution

0.1 V

Accuracy± (1 % of rdng + 2 digits) (10 V to 424 V and at room temperature)

Resolution

0.01 A

Resolution
Resolution
Resolution

0.1 A±(1 % of rdng + 2 digits) (5 % of the max. rated current to max. rated current and at room temperature)

0.1 W / 1W

1W±(1 % of rdng +3 digits) (10 % of the rated power capacity to the rated power capacity, when the load power factor is 1, and at room temperature.)

100 V input type or 200 V input type
When the input voltage is 100 V or 200 V, the output voltage is 100 V or 200 V, the output current is the rated value, the load power factor is 1, and the output frequency is between 40 Hz and 999.9 Hz.
When the output frequency is between 45 Hz and 65 Hz, with no load, and at room temperature.
When the maximum voltage is between 1 V and 100 V (L range) or 2 V and 200 V (H range) and the load power factor is between 0.8 and 1.
When the output voltage is between 100 V and 150 V (L range) or 200 V and 300 V (H range), the output current is reduced by the output voltage.
When the load power factor is between 0 and 0.8, the output current is reduced by the load power factor.
When the output frequency is between 1 Hz and 40 Hz, the output current is reduced by the output frequency.
*5 For capacitor-input rectifier loads (however, this is limited by the rated output current’s rms value).
*6 When the output voltage is 100 V or 200 V and the output frequency is between 40 Hz and 999.9 Hz (reverse current is -180 deg out of phase with the output voltage).
*7 With no load at room temperature*8 When the output voltage is between ±100 V and ±212 V (L range) or ±200 V and ±424 V (H range), the output current is reduced by the output voltage.
*9 Limited by the rated output current’s rms value*10 With respect to changes in the rated range*11 With respect to 0 % to 100 % changes in the rating
When the output voltage is between 80 V and 150 V (L range) or 160 V and 300 V (H range) and the load power factor is 1. At the output terminal block. When the response mode is set to FAST or MEDIUM.
*12 Between 40 Hz and 999.9 Hz.
When the output voltage is between 80 V and 150 V (L range) or 160 V and 300 V (H range) and the load power factor is 1. This is the output line regulation with 200 Hz as the reference.
*13 With respect to changes in the rated range
When the output voltage range is 100 V or 200 V and the output current is 0 A.
*14 With respect to changes in all rated ranges*15 When the output voltage is between 80 V and 150 V (L range) or 160 V and 300 V (H range) and the load power factor is 1.
*16 When the output voltage is 100 V or 200 V, the load power factor is 1, and the output current changes from 0 A to the rated value and from the rated value to 0 A.
*17 When the input voltage is 100 V or 200 V, the output voltage is 100 V or 200 V, the output current is the rated value, the load power factor is 1, and the output frequency is between 40 Hz and 999.9 Hz.
*18 With the true rms display, a waveform with a crest factor of 3 or less, DC, output frequency between 40 Hz and 999.9 Hz, RMS, and AVE.
*19 When the output frequency is between 45 Hz and 65 Hz.

AC Power Supply

PCR-LE series
Item/Model

PCR500LE

PCR1000LE

PCR2000LE

PCR3000LE

PCR4000LE

PCR6000LE

1P2W

BNC terminals

3P3W200V

PCR9000LE
3P4W400V

3P3W200V

3P4W400VSEQ TRIG OUT *1

Pulse width approx. 10 μs, open collector output, pullup at +5 V and approx. 10 kΩ serial resistance approx. 220 Ω, maximum sink current 10 mA, BNC connector

SEQ STAT OUT *1

Step time output, open collector output, pullup at +5 V and approx. 10 kΩ serial resistance approx. 220 Ω, maximum sink current 10 mA, BNC connector

SEQ TRIG IN *1

Operating pulse width 10 μs or greater, photo-coupler input, driving voltage 5 V, serial resistance approx. 470 Ω, active with 7 mA source, BNC connector

Limit Values and Protection Functions

Voltage

AC voltage upper limit
AC voltage lower limit

0.0 V to 305.0 V

DC voltage upper limit
DC voltage lower limit-431.0 V to +431.0 V

Output overvoltage protection
AC/AC+DC mode

0.0 V to 474.1 V

Output overvoltage protection
DC mode-474.1 V to +474.1 V

Output undervoltage protection
AC/AC+DC mode

0.0 V to 474.1 V

Output undervoltage protection
DC mode-474.1 V to +474.1 V

Resolution

0.1 V

Upper limit
Frequency Lower limit

1 Hz to 999.9 Hz *2

Resolution

Current

0.01 Hz (1.00 Hz to 100.0 Hz), 0.1 Hz (100.0 Hz to 999.9 Hz)

Current limit*3 AC mode 0.50 A to 5.50 A 1.00 A to 11.00 A 2.00 A to 22.00 A 3.00 A to 33.00 A 4.00 A to 44.00 A

6.00 A to 66.00 A

Current limit*3 DC/AC+DC mode 0.35 A to 3.85 A 0.70 A to 7.70 A 1.40 A to 15.40 A 2.10 A to 23.10 A 2.80 A to 30.80 A

4.20 A to 46.20 A

6.30 A to 69.30 A

Positive peak current limit*4 0.50 A to 22.00 A 1.00 A to 44.00 A 2.00 A to 88.00 A 3.00 A to 132.0 A 4.00 A to 176.0 A

6.00 A to 264.0 A

9.00 A to 396.0 A

Negative peak current limit*4 -0.50 A to -22.00 A -1.00 A to -44.00 A -2.00 A to -88.00 A -3.00 A to -132.0 A -4.00 A to -176.0 A-6.00 A to -264.0 A-9.00 A to -396.0 A

Resolution*5

9.00 A to 99.00 A

0.01 A (0.35 A to 100.0 A), 0.1 A (100.0 A to 396.0 A)

GeneralBetween input and chassis,
Insulation
output and chassis, andresistance
input and output
Between input and chassis,
Withstandoutput and chassis, andvoltage
input and output
Circuit method
Operatingenvironment
Operatingtemperature range
Storage temperature
Environmental range
conditions
Operating humidity
range
Storage humidity
range
Altitude

500 Vdc, 30 MΩ or more

500 Vdc, 10 MΩ or more

1.5 kVAC for 1 minuteLinear amplifier system
Indoor use, overvoltage category II
0 °C to +50 °C-10 °C to +60 °C
20 % rh to 80 % rh (no condensation)
90 % rh or less (no condensation)
Up to 2000 m
Approx.17 kg(37.4 Ibs)

Approx. 35 kg(77.1 Ibs)

Approx. 55 kg(121.2 Ibs)

Approx. 82 kg(180.7 Ibs)

Approx. 96 kg(211.6 Ibs)

Approx. 140 kg(308.6 Ibs)

Approx. 140 kg(308.6 Ibs)

Approx. 140 kg(308.6 Ibs)

Approx. 190 kg(418.8 Ibs)

Approx. 190 kg(418.8 Ibs)

Input terminal

Inlet

M4

M5

M8

M8

M8

M5

M5

M5

M5

Output terminal

M4

M4

M4

M5

M5

M8

M8

M8

M8

M8

WeightPower cord
Setup guide
Accessories

The input power cable is not included. Please refer to the list of ordering information specified on the last page.
1 copyQuick Reference

1 each for English and Japanese

Safety information

1 copyCD-ROM(User's manual)Electromagnetic compatibility(EMC) *6, 7

Safety *6

1 pc. With plug
Length: 3 m

1 discComplies with the requirements of the following
directive and standards.
Complies with the requirements of the following directive and standards.
EMC Directive 2014/30/EU
EMC Directive 2014/30/EU
EN61326-1(ClassA *8), EN55011(ClassA *8, Group1 *9)
EN61326-1(ClassA *8), EN55011(ClassA *8, Group1 *9)
EN61000-3-2 *10, EN61000-3-3 *10
The maximum length of all cables and wires connected to the PCR-LE Series must be less than 3 m.
The maximum length of all cables and wires
connected to the PCR-LE Series must be less than 3 m.
Complies with the requirements of the following directive and standard.
Low Voltage Directive 2014/35/EU *7
EN 61010-1
Class I *11, Pollution Degree 2

*1
*2
*3
*4
*5
*6
*7
*8

Although signals are insulated with output terminals, each signal is common. Logic setting is also possible.
The frequency is limited to the range from 1 Hz to 500.0 Hz when the 3P05-PCR-LE(500HZ LMT) is installed in the PCR-LE series.
The current that can actually be supplied is 1.1 times the rated current or the current limit, whichever is less.
The current that can actually be supplied is the maximum peak current or the current limit, whichever is less.
You can set the current in 0.01 A/ 0.1 A steps, but it may not change at this resolution depending on the relationship with the internal D/A resolution.
Does not apply to specially ordered or modified PCR-LEs.
Only on models that have the CE marking on the panel.
This is a Class A equipment. This product is intended for use in an industrial environment.
This product may cause interference if used in residential areas. Such use must be avoided unless the user takes special measures to reduce electromagnetic emissions to prevent interference to thereception of radio and television broadcasts.
*9 This is a Group 1 equipment. This product does not generate and/or use intentionally radio-frequency energy, in the form of electromagnetic radiation, inductive and/or capacitive coupling,
for the treatment of material or inspection/analysis purpose.
*10 PCR500LE, PCR1000LE, PCR2000LE only.
*11 This is a Class I equipment. Be sure to ground this product’s protective conductor terminal. The safety of this product is only guaranteed when the product is properly grounded.

25

Output single-phase, single-phase 3-wire,*
Convenient multiple output supports a wide
AC power supply offering superior space factor

High-performance AC Power Supplies

PCR-LE2 SERIESThe PCR-LE2 Series are designed based on the PCR-LE Series that supports single-phase output, single-phase 3-wire output *, andthree-phase output within the rated capacity by selecting the switch from the front panel operation. The PCR-LE2 series offer the same
basic performance, using the common power unit of the PCR-LE Series, with providing easier installation and saving the space more
efficiently compare to the individual
allocation of the system for a singlephase, single-phase 3-wire, and threephase systems. The lineup of PCRLE2 Series are available in 5 models:Single-phase output display screen

Single-phase 3-wire output display screen Three phase output display screen

6 kVA, 9 kVA, 12 kVA, 18 kVA, and27 kVA model.
*2/3 of the rated output power

PCR6000LE2

26

PCR9000LE2

PCR27000LE2

and three-phase power with a single unit.
range of industrial devices.
and cost performance.

AC Power Supply

PCR-LE2 series*: The Output power with single-phase 3-wire limits 2/3 of the rated output.

● Lineup
PCR6000LE2

PCR9000LE2

PCR12000LE2

PCR18000LE2

PCR27000LE2

Single-phase,Three phase 4-wire

6 kVA

9 kVA

12 kVA

18 kVA

27 kVA

Single phase 3-wire

4 kVA

6 kVA

9 kVA

12 kVA

18 kVA

60 A / 30 A

90 A / 45 A

120 A / 60 A

180 A / 90 A

270 A / 135 A

20 A / 10 A

30 A / 10 A

40 A / 20 A

60 A / 30 A

90 A / 45 A

Model
Outputcapacity

Single-phase
Maximum
output current Single phase 3-wire

1 V to 150 V / 2 V to 300 V
ACmode(L/H range)

Single-phase

60 A / 30 A

90 A / 45 A

120 A / 60 A

180 A / 90 A

270 A / 135 A

Three phase 4-wire

20 A / 10A

30 A / 15 A

40 A / 20 A

60 A / 30 A

90 A / 45 A

±1.4 V to ±212 V / ±2.8 V to ±424 V
DC mode(L/H range)

Single-phase

42 A / 21 A

63 A / 31.5 A

84 A / 42 A

126 A / 63 A

189 A / 94.5 A

Single phase 3-wire

14 A / 7A

21 A / 10.5 A

28 A / 14 A

42 A / 21 A

63 A / 31.5 A

430 (16.93")(445 (17.52")) W

430 (16.93")(445 (17.52")) W(1585 (62.40")) W(1585 (62.40")) W(1585 (62.40")) W

944 (36.17")(1040 (40.94")) H

1325 (52.17")(1420 (55.91")) H(790 (31.10")) H(1045 (41.14")) H(1425 (56.10")) H

550 (21.65")(595 (23.43")) D

550 (21.65")(595 (23.43")) D(835 (32.87")) D(835 (32.87")) D(835 (32.87")) D

Approx. 140 kg(308.6 Ibs)

Approx. 190 kg(418.8 Ibs)

Approx. 350 kg(771.6 Ibs)

Approx. 480 kg(1058.2 Ibs)

Approx. 630 kg(1388.9 Ibs)

Dimensions (mm(inches))(Maximum dimensions)

Weight*OP03-KRC included.

*OP03-KRC included.

*OP03-KRC included.

● Rear panel

PCR6000LE2

PCR9000LE2

PCR27000LE2

27

dimensions

440 (17.32)

16 M6 screw holes,
max. depth 10 mm(0.39 ”)

(1.97)
50

430 (16.93)

342 (13.46)

MAX 445 (17.52)

50(1.97)

59.2(2.33)

Unit:mm(inches)440 (17.32)

59.2(2.33)

61.2(2.41)

A-arrow caster screw position diagramMAX 20(0.79)

550 (21.65)
MAX 595 (23.43)
28 M4 screw holes, max. depth 10 mm (0.39 ”)

195 (7.68)

342 (13.46)

MAX 445 (17.52)
430 (16.93)

50(1.97)

50(1.97)

16 M6 screw holes,
max. depth 10 mm (0.39 ”)

550 (21.65)

195 (7.68)

MAX 595 (23.43)

195 (7.68)

257.5 (10.14)

M12 screw hole, max. depth 35 mm (1.38 ”)

A arrow
140.2 40
(5.52) (1.57)

26 (1.02)

200 (7.87)

40(1.57)

22(0.87)

76.5(3.01)

76.5
22 (3.01)(0.87)

195 (7.68)

250 (9.84)

MAX1040 (40.94)944 (36.17)

250 (9.84)

MAX 1420 (55.91)
1325 (52.17)

16 M4 screw holes, max. depth 10 mm (0.39 ”)

8 M10 screw holes,
max. depth
10 mm (0.39 ”)

M12 screw hole,max. depth 35 mm (1.38 ”)

508.5 (20.02)

PCR9000LE2

MAX 835 (32.87)

70(2.76)

MAX 180(7.09)

240 (9.45)

PCR6000LE2

1497 (58.94)
MAX 1585 (62.40)

MAX 1485 (58.46)

MAX 790 (31.10)

MAX 1405 (55.31)

PCR12000LE2
28

195 (7.68)

MAX 20(0.79)

A-arrow caster screw position diagram

195 (7.68)

61.2(2.41)！ Please use caution when installing or moving the system.

！

26(1.02)

A arrow
140.2(5.52)

40(1.57)

200 (7.87)

40(1.57)
508.5 (20.02)

8 M10 screw holes,
max. depth
10 mm (0.39 ”)

Concerning installation &relocation PCR12000LE2,
18000LE2, 27000LE2●The PCR12000LE2,18000LE2,27000LE2 requires for the
installation work. Please consult with your local Kikusui
distributor.
●The PCR12000LE2,18000LE2,27000LE2 cannot be relocated after it is installed. If relocation becomes necessary,please consult with your local Kikusui distributor.

AC Power Supply

MAX 835 (32.87)

70(2.76)

MAX 180(7.09)

240 (9.45)

PCR-LE2 series

1497 (58.94)
MAX 1585 (62.40)

MAX 1485 (58.46)

MAX 1045 (41.14)

MAX 1405 (55.31)

MAX 835 (32.87)！ Please use caution when installing or moving the system.

70(2.76)

MAX 180(7.09)

240 (9.45)

PCR18000LE2

1497 (58.94)
MAX 1585 (62.40)

MAX 1485 (58.46)

MAX 1425 (56.10)

MAX 1405 (55.31)

PCR27000LE2！ Please use caution when installing or moving the system.

29

specifications
Item/Model

PCR6000LE2
1P2W

Input ratings (AC rms)

PCR9000LE2

3P3W200V
Line voltage
170 V to 250 V

VoltagePhasesSingle phase

Three phase 3-wire

Frequency

3P4W400V

3P3W200V

3P4W400V

Line voltage 324 V to 440 V

Line voltage 324 V to 440 V(Phase voltage 187 V to 254 V)

Line voltage
170 V to 250 V(Phase voltage 187 V to 254 V)

Three phase 4-wireThree phase 3-wireThree phase 4-wire

47 Hz to 63 Hz

Apparent power

Approx. 10.6 kVA

Approx. 15.7 kVA

Power factor *1

0.97 (TYP)

Max. current

64 A or less

38 A or less

21 A or less

55 A or less

30 A or less

AC mode output ratings (AC rms)
Phase voltage (output L range, output H range)*2

1 V to 150 V / 2 V to 300 V

Voltage setting range

0 V to 152.5 V / 0 V to 305.0 V

Voltage setting accuracy (output L range, output H range)*3
Max. current*4±(0.3 % of set + 0.6 V)

Single phase, poly phase, L range, H range

60 A, 30 A · 20 A, 10 A

Phase*5

90 A, 45 A · 30 A, 15 A

Single phase · Single phase3-wire · Three phase 4-wirePower capacity

Single phase, Three-phase 4-wire, Single phase 3-wire

6 kVA · 4 kVA

9 kVA · 6 kVA

Maximum peak current*6

Max. current (rms) × 4 (TYP)

Max. reverse current*7

30 % of the max. current (rms)

Load power factor*4

0 to 1 (leading or lagging)

Frequency*4 *8 *9

1 Hz to 999.9 Hz ★

DC mode output ratings, AC+DC mode (for Single-phase and Single-phase Three-wire output only)
Phase voltage (output L range, output H range)*2±1.4 V to ±212 V / ±2.8 V to ±424 V

Voltage setting range-215.5 V to +215.5 V / -431.0 V to +431.0 V

Voltage setting accuracy (output L range, output H range) *10
Max. current*4± (0.05 % of set + 0.05 V / 0.1 V)

Single phase, Single phase 3-wire and Three-phase, L range, H range

42 A, 21 A · 14 A, 7 A

63 A, 31.5 A · 21 A, 10.5 A

Max. instantaneous current*11
Power capacity

Max. current (rms) × 3.6

Single phase, Single phase 3-wire, Three-phase

4.2 kW · 2.8 kW

6.3 kW · 4.2 kW

Output voltage stability
Line regulation(With respect to changes in the rated range)

Within ±0.1 %Line regulation(With respect to 0 % to 100 % changes in the rating)*12±0.3 V

Output frequency variation in AC mode(Between 40 Hz and 999.9 Hz)*13

Within ±0.5 %Ripple noise in DC mode(5 Hz to 1 MHz components)

0.25 Vrms or lessAmbient temperature variation(With respect to changes in the rated range)*14

100 ppm/ °C (TYP)

Output frequency stability, output voltage waveform distortion ratio, output voltage response speed, efficiency
Within ±5×10 - 5, Setting accuracy : Within ±1×10 - 4

Output frequency stability(With respect to changes in all rated ranges)
Output voltage waveform distortion ratio*15

0.3 % or less

Output voltage response speed*16

30 µs (TYP)

Efficiency*1

58 % or more

Phase difference of theoutput phase voltage *17

Resolution

1 deg
Within ± (0.4° + f0×1.8×10 - 3) deg

Accuracy

f0 is the output frequency *18

Meters (fluorescent display)
Voltmeter*19 *20

Resolution RMS,AVE Display mode

0.1 V

Accuracy RMS,AVE Display mode

Within ± (1 % of rdng + 2 digits) (10 V to 848 V and at room temperature)

Ammeter*19 *20

Resolution RMS,AVE Display mode Single phase · Poly phase

Wattmeter*20

0.1A · 0.01 A

Accuracy RMS Display mode
Resolution Single phase · Poly phase
AccuracyFrequency meter*21

0.1 A

Within ± (1% of reading + 2digits) (5 % of the max. rated current to max. rated current and at room temperature)
1 W · 0.1 W / 1 W

1W

Within ± (1 % of reading + 3digits) (10 % of the rated power capacity to the rated power capacity, when the load power factor is 1, and at room temperature.)
Resolution

0.01 Hz / 0.1 Hz

Between input and chassis, output andchassis, and input and output

500 V, 10 MΩ or more

GeneralInsulation resistanceWithstand voltage

1.5 kVAC for 1 minuteCircuit method
Environmental
conditions

Linear amplifier system
Operating temperature range / Storage temperature range

0 °C to +50 °C / -10 °C to +60 °C

Operating humidity range / Storage humidity range

20 % rh to 80 % rh (no condensation) / 90 % rh or less (no condensation)

WeightApprox.140 kg (308.6 Ibs)

Input terminal

Input terminal board [3φ]Output terminal

Output terminal board Single phase · Single
phase 3-wire,Three-phase 4-wire

Input power cord
[Sold separately option]

M8

Conductor cross section/Length

single-core cable
3 pc

4 pc

5 pc

4 pc

5 pc

14 mm2 / 3 m

8 mm2 / 3 m

5.5 mm2 / 3 m

14 mm2 / 3 m

5.5 mm2 / 3 m

Setup guide
Accessories

1 copyQuick Reference

1 each for English and Japanese

Safety information

1 copyCD-ROM(User's manual)
Other
*1
*2
*3
*4

*5
*6
*7
*8
*9
*10
*11
*12

30

M5
M8 · M5Shape
The number

Approx.190kg (418.8 Ibs)
M5

Electromagnetic compatibility (EMC)
Safety1 disc
EMC Directive 2014/30/EU. EN61326-1. EN 55011 The maximum length of all cables and wires connected to the PCR-LE Series must be less than 3 m.

Low Voltage Directive 2014/35/EU, EN61010-1 (Class I, Pollution Degree2)

When the output phase voltage is 100 V or 200 V, the output current is the rated value, the load power factor is 1, and theoutput frequency is between 40 Hz and 999.9 Hz.
L/H range can be changed by means of a switch on the front panel. Resolution: 0.1V
When the output frequency is between 45 Hz and 65 Hz, with no load, and at room temperature.
When the maximum voltage is between 1 V and 100 V (L range) or 2 V and 200 V (H range) and the load power factor is
between 0.8 and 1.When the output phase voltage is between 100 V and 150 V or 200 V and 300 V (AC mode) or ±100 V
and ±212 V or ±200 V and ±424 V (DC mode), the output current is reduced by the output phase voltage.
When the load power factor is between 0 and 0.8, the output current is reduced by the load power factor. (AC mode)
When the output frequency is between 1 Hz and 40 Hz, the output current is reduced by the output frequency. (AC mode)
The output phase mode can be changed by means of a key on the operation panel. "Poly" in the table indicates singlephase three-wire mode and three-phase four-wire mode.
When the output phase voltage is in the vicinity of the peak (±15 deg) (However, this is limited by the rated output current’s rms value).
When the output phase voltage is 100 V or 200 V and the output frequency is between 40 Hz and 999.9 Hz (reverse current
is −90 deg to −180 deg / 90 deg to 180 deg out of phase with the output voltage).
Resolution : 0.01Hz (1.00 Hz to 100.0 Hz), 0.1Hz(100.0 Hz to 999.9 Hz)
The "500Hz Limit Model" limits the maximum frequency up to 500Hz under the "Three-phase output".
With no load at room temperature
Limited by the rated output current's rms value
When the output phase voltage is between 80 V and 150 V (L range) or 160 V and 300 V (H range) and the load power
factor is 1. At the output terminal block. When the response mode is set to MEDIUM.(There is no F mode)

*13

*14
*15
*16
*17
*18

*19
*20
*21

When the output phase voltage is between 80 V and 150 V (L range) or 160 V and 300 V (H range) and the load
power factor is 1. This is the output line regulation with 200 Hz as the reference. When the response mode is set
to MEDIUM.(There is no F mode)
When the output phase voltage is 100 V or 200 V and the output current is 0 A.
When the output phase voltage is between 80 V and 150 V (L range) or 160 V and 300 V (H range) and the load
power factor is 1. When the response mode is set to MEDIUM.(There is no F mode)
When the output phase voltage is 100 V or 200 V, the load power factor is 1, and the output current changes
from 0 A to the rated value and from the rated value to 0 A.
Phase difference between output voltages (phase voltages) when each phase is considered along with theneutral point.
The following show the angles obtained by calculating the expression with the specified frequency.
When phase difference is 120 deg.
Within 120 ± 0.5 deg(when generating 60 Hz output)
Within 120 ± 1.2 deg(when generating 400 Hz output)
With the true rms display, a waveform with a crest factor of 3 or less.
When the output frequency is between 45 Hz and 65 Hz.
Displays the output frequency setting (frequency of the internal reference voltage)★ PCR-LE2 Series 500Hz Limit Model

The PCR-LE Series offers the type on each model that limits the maximum output frequency up to 500 Hz.

AC Power Supply

PCR-LE2 series
PCR12000LE2

PCR18000LE2

PCR27000LE2

3P3W200V

3P4W400V

3P3W200V

3P4W400V

3P3W200V

3P4W400V

Line voltage
170 V to 250 V

Line voltage 324 V to 440 V

Line voltage 324 V to 440 V(Phase voltage 187 V to 254 V)

Line voltage
170 V to 250 V

Line voltage 324 V to 440 V(Phase voltage 187 V to 254 V)

Line voltage
170 V to 250 V(Phase voltage 187 V to 254 V)

Three phase 3-wireThree phase 4-wireThree phase 3-wireThree phase 4-wireThree phase 3-wireThree phase 4-wire

47 Hz to 63 Hz
Approx. 23 kVA

Approx. 33 kVA

Approx. 48 kVA

0.97 (TYP)
75 A or less

39 A or less

111 A or less

59 A or less

165 A or less

91 A or less

1 V to 150 V / 2 V to 300 V
0 V to 152.5 V / 0 V to 305.0 V±(0.3 % of set + 0.6 V)
120 A, 60 A · 40 A, 20 A

180 A, 90 A · 60 A, 30 A

270 A, 135 A · 90 A, 45 A

Single phase · Single phase 3-wire · Three phase 4-wire
12 kVA · 8 kVA

18 kVA · 12 kVA

27 kVA · 18 kVA

Max. current (rms) × 4 (TYP)
30 % of the max. current (rms)
0 to 1 (leading or lagging)
1 Hz to 999.9 Hz ★
±1.4 V to ±212 V / ±2.8 V to ±424 V-215.5 V to +215.5 V / -431.0 V to +431.0 V± (0.05 % of set + 0.05 V / 0.1 V)
84 A, 42 A · 28 A, 14 A

126 A, 63 A · 42 A, 21 A

189 A, 94.5 A · 63 A, 31.5 A

Max. current (rms) × 3.6
8.4 kW · 5.6 kW

12.6 kW · 8.4 kW

18.9 kW · 12.6 kW

Within ±0.1 %±0.5 V
Within ±1 %0.5 Vrms or less
100 ppm/ °C (TYP)
Within ±5×10 - 5, Setting accuracy : Within ±1×10 - 4
0.5 % or less
50 µs (TYP)
58 % or more
1 degWithin ± (0.4° + f0 × 1.8 × 10 - 3) deg

f0 is the output frequency *18

0.1 V
Within ± (1 % of rdng + 2 digits) (10 V to 848 V and at room temperature)
0.1 A

0.1 A / 1 A · 0.1 A

Within ± (1% of reading + 2digits) (5 % of the max. rated current to max. rated current and at room temperature)
1 W / 10 W
Within ± (1 % of reading + 3digits) (10 % of the rated power capacity to the rated power capacity, when the load power factor is 1, and at room temperature.)
0.01 Hz / 0.1 Hz
500 V, 10 MΩ or more
1.5 kVAC for 1 minuteLinear amplifier system
0 °C to +50 °C / -10 °C to +60 °C
20 % rh to 80 % rh (no condensation) / 90 % rh or less (no condensation)
Approx.350 kg (771.6 Ibs)

Approx.480 kg (1058.2 Ibs)

Approx.630 kg (1388.9 Ibs)

M8

M8

M8

M8 · M8

Required for the installation work, contact local distributor.
1 copy1 each for English and Japanese
1 copy1 disc—IEC 61010-1 (Class I, Pollution Degree2)

31

specifications
PCR6000LE2

Item/Model

PCR9000LE2

PCR12000LE2

PCR18000LE2

PCR27000LE2

Limit Values and Protection Functions

Voltage

AC voltage upper limit
AC voltage lower limit

0.0 V to 305.0 V

DC voltage upper limit
DC voltage lower limit-431.0 V to + 431.0 V

Output overvoltage protection
AC/AC+DC mode

0.0 V to 474.1 V

Output overvoltage protection
DC mode-474.1 V to + 474.1 V

Output undervoltage protection
AC/AC+DC mode

0.0 V to 474.1 V

Output undervoltage protection
DC mode-474.1 V to + 474.1 V

Resolution

0.1 V

Upper limit
Frequency Lower limit

1 Hz to 999.9 Hz, 500 Hz LMT model: 1 Hz to 500 Hz (Three-phase output)

Resolution
Current limit *1
AC modeCurrent limit *1
DC/AC+DC mode
Current

Positive peakcurrent limit *2
Negative peakcurrent limit *2
Resolution *3

Single-phase output

6.00 A to 66.00 A

0.01 Hz (1.00 Hz to 100.0 Hz), 0.1 Hz (100.0 Hz to 999.9 Hz)
9.00 A to 99.00 A
12.00 A to 132.0 A
18.00 A to 198.0 A

Single-phase three-wire output
Three-phase output

2.00 A to 22.00A

3.00 A to 33.00 A

4.00 A to 44.00 A

6.00 A to 66.00 A

9.00 A to 99.00 A

Single-phase output

4.20A to 46.20A

6.30 A to 69.30 A

8.40 A to 92.40 A

12.60 A to 138.6 A

18.90 A to 207.9 A

Single-phase three-wire output
Three-phase output

1.40A to 15.40A

2.10 A to 23.10 A

2.80 A to 30.80 A

4.20 A to 46.20 A

6.30 A to 69.30 A

Single-phase output

6.00A to 264.0A

9.00 A to 396.0 A

12.00 A to 528.0 A

18.00 A to 792.0 A

27.00 A to 1188 A

Single-phase three-wire output
Three-phase output

2.00A to 88.00A

3.00 A to 132.0 A

4.00 A to 176.0 A

6.00 A to 264.0 A

9.00 A to 396.0 A

Single-phase output-6.00A to -264.0A-9.00 A to -396.0 A-12.00 A to -528.0 A-18.00 A to -792.0 A-27.00 A to -1188 A

Single-phase three-wire output
Three-phase output-2.00A to -88.00A-3.00 A to -132.0 A-4.00 A to -176.0 A-6.00 A to -264.0 A-9.00 A to -396.0 A

0.01 A (0.35 A to 100.0 A) , 0.1A (100.0 A to 1000 A) , 1 A (1000 A to 1188 A)*1 The current that can actually be supplied is 1.1 times the rated current or the current limit, whichever is less.
*2 The current that can actually be supplied is the maximum peak current or the current limit, whichever is less.
*3 You can set the current in 0.01 A/ 0.1 A/ 1 A steps, but it may not change at this resolution depending on the relationship with the internal D/A resolution.

32

27.00 A to 297.0 A

common specifications■ Rated output current characteristics (Derating)
The output voltage ratio is a percentage where 100 % represents an output voltage of 100 V (output L range) or 200 V (output
H range) in AC mode or DC mode.
The output current ratio is a percentage where 100 % represents the maximum rated output current in AC mode or DC mode.

Load power factor versus rated
output current characteristics

Output voltage ratio versus rated output
current characteristics (AC mode)100

AC modeOutput current ratio [%]Output current ratio [%]

100
80
60
40
20
0

20

40 60 80 100 120 140150
Output voltage ratio [%]100

The second quadrant

40
20

40
20

40

80

120 160 220-40 -20
-40

[ ]

%

-60
-80
The third quadrant

1

100

60

00

0.8 0.6 0.4 0.2 0
0.2 0.4 0.6 0.8
Tidal current
Reverse current
Load power factor

The first quadrant

80-220 -160 -120 -80

1Output frequency versus rated
output current characteristics-100

Outputvoltage ratio
出力電圧率[%][%]

The fourth quadrant

Output current ratio [%]Output current ratio [%]

60

0

0

Output voltage ratio versus rated output
current characteristics (DC mode)

率

80

80
60
40
20
0

1

10

20 30 40 50
Output frequency [Hz]999.9

For the “Output voltage ratio versus rated output current characteristics (AC mode)” and “Load power factor versus rated output
current characteristics” graphs, the rated output current is the product of the output current ratios shown in both graphs. The output
current ratio shown in the “Output frequency versus rated output current characteristics” graph is given priority if it is less than the
product of the output current ratios described above. (This only applies to AC mode.)33

High-performance multifunctional
PCR-LE/ LE2 SERIES SELECTION■ Extended system for large capacity applications. Flexible configuration in models.
PCR-LE2

PCR-LE
0.5

1K

2K

3K

4K

6K

Max. output capacity

9K

0.5

1K

2K

6K

9K

3K

4K

12K

18K

6K

27K

10K

9K

12K

PCR-LE

54K

27K

18K

81K

★
Flexible
configurationin models.

Single-phase

To 27 kVA•••••••••••••■ Parallel operation : 2 kVA or higher model / Max. connectable units: 5 / Max. expansion ★Combinations of different models are possible!

PCR-LE

Flexible
configurationin models.

Single phase3-wire★

To 54 kVA•••••••••••■ Single phase 3-wire operation : All models ★Combinations of different models are possible!

PCR-LE

Flexible
configurationin models.

★

Three-phase•••••■ Three-phase operation : All models ★Combinations of different models are possible!

PCR-LE2

Single phase3-wireTo 18 kVA*The Output power with single-phase 3-wire limits 2/3 of the rated output.

PCR-LE2
Single-phase
Three-phase

To 27 kVA

Ordering information The system configuration (Model and Options)
Model
Part

High-performance
AC Power Supplies(Single phase)

High-performance ACPower Supplies(Single phase /Single phase three wire /Three-phase switchable
type)

34

Model

Dimensions (Maximum dimensions)

WeightPower cable

PCR500LE
PCR1000LE
PCR2000LE
PCR3000LE
PCR4000LE
PCR6000LE
PCR6000LE (3P3W 200V)
PCR6000LE (3P4W 400V)
PCR9000LE (3P3W 200V)
PCR9000LE (3P4W 400V)
PCR6000LE2
PCR6000LE2 (3P3W 200V)
PCR6000LE2 (3P4W 400V)
PCR9000LE2 (3P3W 200V)
PCR9000LE2 (3P4W 400V)
PCR12000LE2 (3P3W 200V)
PCR12000LE2 (3P4W 400V)
PCR18000LE2 (3P3W 200V)
PCR18000LE2 (3P4W 400V)
PCR27000LE2 (3P3W 200V)
PCR27000LE2 (3P4W 400V)(430(16.93"))W×173(6.81"(195(7.68")))H×550(21.65(600(23.62")))Dmm(430)W×262(345)H×550(595)Dmm(430)W×389(475)H×550(595)Dmm
430(445)W×690(785)H×550(595)Dmm
430(445)W×690(785)H×550(595)Dmm
430(445)W×944(1040)H×550(595)Dmm
430(445)W×944(1040)H×550(595)Dmm
430(445)W×944(1040)H×550(595)Dmm
430(445)W×1325(1420)H×550(595)Dmm
430(445)W×1325(1420)H×550(595)Dmm
430(445)W×944(1040)H×550(595)Dmm
430(445)W×944(1040)H×550(595)Dmm
430(445)W×944(1040)H×550(595)Dmm
430(445)W×1325(1420)H×550(595)Dmm
430(445)W×1325(1420)H×550(595)Dmm(1585)W×(790)H×(835)Dmm(1585)W×(790)H×(835)Dmm(1585)W×(1045)H×(835)Dmm(1585)W×(1045)H×(835)Dmm(1585)W×(1425)H×(835)Dmm(1585)W×(1425)H×(835)Dmm

Approx. 17 kg(37.4 lbs)
Approx. 35 kg(77.1 lbs)
Approx. 55 kg(121.2 lbs)
Approx. 82 kg(180.7 lbs)
Approx. 96 kg(211.6 lbs)
Approx. 140 kg (308.6 Ibs)
Approx. 140 kg (308.6 Ibs)
Approx. 140 kg (308.6 Ibs)
Approx. 190 kg(418.8 Ibs)
Approx. 190 kg(418.8 Ibs)
Approx. 140 kg (308.6 Ibs)
Approx. 140 kg (308.6 Ibs)
Approx. 140 kg (308.6 Ibs)
Approx. 190 kg(418.8 Ibs)
Approx. 190 kg(418.8 Ibs)
Approx. 350 kg(771.6 Ibs)
Approx. 350 kg(771.6 Ibs)
Approx. 480 kg(1058.2 Ibs)
Approx. 480 kg(1058.2 Ibs)
Approx. 630 kg(1388.9 Ibs)
Approx. 630 kg(1388.9 Ibs)

Included as a standard accessory
AC5.5-3P3M-M4C
AC8-1P3M-M5C-3S
AC14-1P3M-M8C-3S
AC22-1P3M-M8C-3S
AC14-1P3M-M8C-3S
AC14-1P3M-M5C-4S
AC5.5-1P3M-M5C-5S
AC14-1P3M-M5C-4S
AC5.5-1P3M-M5C-5S
AC14-1P3M-M8C-3S
AC14-1P3M-M5C-4S
AC5.5-1P3M-M5C-5S
AC14-1P3M-M5C-4S
AC5.5-1P3M-M5C-5S*Load cable is not included. Please make prior arrangements or consult your local Kikusui distributor.

Included in the installation fee.
(The installation fee is required as anadditional cost)*Please consult with your local distributor.

AC Power Supply

AC Power Supplies
GUIDE

PCR-LE2 series

New stage of AC power supply supporting new energy field

The PCR-LE Series is a high performance and multifunctional
AC power supply. It can be used as a high quality and stability of
the regulated power supply and it controls the waveform freely of
the broadband frequency by taking the advantage characteristicsof the linear amplifier method. Furthermore, it supports the low
frequency immunity test and various power environment tests
combined with various options. The options are available for the
Parallel Operation, Single-phase Three-wires Operation, andThree-phase Operation that enables you to expand the system
for the Single-phase Operation up to 27 kVA, Single-phase Threewires Operation up to 54 kVA, and Three-phase Operation up to
81kVA for which systems can be applied to the large-scale EMC
testing site. The PCR-LE Series are available in total of 7 models
for 0.5 kVA, 1 kVA, 2 kVA, 3 kVA, 4 kVA, 6 kVA, and 9 kVA model.

The PCR-LE2 Series are designed based on the PCR-LE Series
that supports single-phase output, single-phase 3-wire output *,and three-phase output within the rated capacity by selecting theswitch from the front panel operation. The PCR-LE2 series offer
the same basic performance, using the common power unit of the
PCR-LE Series, with providing easier installation and saving thespace more efficiently compare to the individual allocation of thesystem for a single-phase, single-phase 3-wire, and three-phasesystems. The lineup of PCR-LE2 Series are available in 3 models:
6 kVA, 9 kVA, 12 kVA, 18 kVA, and 27 kVA model.
*2/3 of the rated output power

PCR-LE Series

PCR-LE2 Series■High-quality/high-stability output with a high-speed linear amp■Capable of various power line abnormality simulations
and the sequence operation■Single phase 500 VA to 9 kVA, supporting the system for the
single-phase, and expandable with optional drivers for the
single-phase three-wire, and three-phase operation■Expandable capacity up to 27 kVA (single-phase),
54 kVA (single-phase three-wires), and 81 kVA (three-phase)■Equipped with various measuring functions■Features a full range of measuring functions and supports
AC, DC, and AC + DC Outputs■Detachable front panel■Eco-friendly function equipped■High-quality/high-stability output with a high-speed linear amp■Capable of various power line abnormality simulations andthe sequence operation■Single-phase 6 kVA to 27 kVA, Capable of the Single-phase output,
Single-phase 3-wire output, and Three-phase output.
■Equipped with various measuring functions■Features a full range of measuring functions and supports AC, DC,
and AC + DC Outputs■Detachable front panel■Eco-friendly function equipped

To 81 kVA

*Applied to
108 kVA, 135 kVA!
*Subject to the costom products*Please consult us for system
power exceeding 108 kVA.

OptionParallel operation
driverSingle-phase three-wire Three-phase output driver Extension cable
Extension
output driverconnection cable

Extension power Power-sync cablesignal cable

3P05-PCR-LE
2P05-PCR-LE

PD05S-PCR-LE(Slave)3P05-PCR-LE(500Hz LMT)
*Overseas export

CC01-PCR-LE(1.5m)
CC02-PCR-LE(2.8m)*2P05/3P05

Interface

Analogcontrol panel

KRB4
KRB200(PCR500LE)–

PD05M-PCR-LE(Master)

Rack mountPC01-PCR-LE(1.3m)

CC11-PCR-LE(1m)

LC01-PCR-LE
(1m)

KRB6
KRB300(PCR1000LE)
KRB9
KRB400-PCR-LE(PCR2000LE)

IB05-PCR-LE(GPIB Interface)
US05-PCR-LE(USB Interface)EX05-PCR-LE
EX06-PCR-LE

EC05-PCR(2m)

LN05-PCR-LE(LAN Interface)
–

–

–

–

–

–

–

–

*Single-phase
operation only for the
PCR6000LE2,
PCR9000LE2*Any one of the following *Any one of the following
can be installed.
can be installed.

35

ordering informationPart

High-performance AC Power Supplies (Single phase)

High-performance AC Power Supplies(Single phase/Single phase three wire/Three-phase switchable type)

Model

Remarks

PCR500LE

Single phase 500 VA

PCR1000LE

Single phase 1 kVA

PCR2000LE

Single phase 2 kVA

PCR3000LE

Single phase 3 kVA

PCR4000LE

Single phase 4 kVA

PCR6000LE

Single phase 6 kVA

PCR9000LE

Single phase 9 kVA

PCR6000LE2

Single phase / Three-phase 6 kVA, Single phase three wire 4 kVA

PCR9000LE2

Single phase / Three-phase 9 kVA, Single phase three wire 6 kVA

PCR12000LE2

Single phase / Three-phase 12 kVA, Single phase three wire 8 kVA

PCR18000LE2

Single phase / Three-phase 18 kVA, Single phase three wire 12 kVA

PCR27000LE2

Single phase / Three-phase 27 kVA, Single phase three wire 18 kVA

GPIB interface

IB05-PCR-LE

USB interface

US05-PCR-LE

LAN interface

LN05-PCR-LE
EX05-PCR-LE

An amplifier typeEX06-PCR-LE

Amplitude control type

For PCR1000LE

AC5.5-3P3M-M4C

3-core cabtire cables 5.5 mm2/3 m M4

For PCR2000LE

AC8-1P3M-M5C-3S

3 single-core cables 8 mm2/3 m M5

For PCR3000LE/6000LE

AC14-1P3M-M8C-3S

3 single-core cables 14 mm2/3 m M8

For PCR4000LE
AC22-1P3M-M8C-3S
Input power
For PCR6000LE (Three-phase 200V)/9000LE (Three-phase 200V) AC14-1P3M-M5C-4S
cable
For PCR6000LE (Three-phase 400V)/9000LE (Three-phase 400V) AC5.5-1P3M-M5C-5S

3 single-core cables 22 mm2/3 m M8

Analog interface

4 single-core cables 14 mm2/3 m M5
5 single-core cables 5.5 mm2/3 m M5

AC14-1P3M-M8C-3S

3 single-core cables 14 mm2/3 m M8

For PCR6000LE2 (Three-phase 200V)/9000LE2 (Three-phase 200V) AC14-1P3M-M5C-4S

4 single-core cables 14 mm2/3 m M5

For PCR6000LE2

For PCR6000LE2 (Three-phase 400V)/9000LE2 (Three-phase 400V) AC5.5-1P3M-M5C-5S

5 single-core cables 5.5 mm2/3 m M5

Extension cable for control panel

EC05-PCR

2m

Parallel operation driver (Master)

PD05M-PCR-LE

Cannot be used with PCR500LE or PCR1000LE.

Parallel operation driver (Slave)

PD05S-PCR-LE

Cannot be used with PCR500LE or PCR1000LE.

Single-phase three-wire output driver

2P05-PCR-LE

Three-phase output driver

3P05-PCR-LE
3P05-PCR-LE (500 Hz LMT)

Overseas export

CC01-PCR-LE

For 2P05 and 3P05, 1.5 m

CC02-PCR-LE

For 2P05 and 3P05, 2.8 m

Extension connection cable (For parallel operation)

PC01-PCR-LE

1.3 m

Extension power signal cable (For parallel operation)

CC11-PCR-LE

1m

Power-sync cable

LC01-PCR-LE

1m

KRB4

For EIA inch size

KRB200For JIS metric size

KRB6

For EIA inch size

KRB300For JIS metric size

KRB9

For EIA inch size

Extension cable

For PCR500LE
Rack mount
Brakets

For PCR1000LE
For PCR2000LE

Base holding angle

IEC dip simulator

KRB400-PCR-LE

For JIS metric sizeOP03-KRC

For fixing PCR3000LE/4000LE/6000LE/9000LE/6000LE2/9000LE2 to the floor.
Standard accessories for the PCR12000LE2/PCR18000LE2/PCR27000LE2.

DSI1020

Single phase 20 A

DSI3020

Single phase / Three-phase 20 A

USB
GPIB

Line impedance network

LIN1020JF

Single phase 20 A

LIN3020JF

Single phase / Three-phase 20 A

LIN3060J

Single phase / Three-phase 60 A exclusive for the JIS standard

OP01-LIN1020JF

LIN1020JF for the "Three-phase" expansion

Quick Immunity Sequencer 2SD009-PCR-LE/WE

Software for creating sequences

SD011-PCR-LE (Wavy for PCR-LE)

Avionics Test Software

SD012-PCR-LE/WE

Remote control software for the Windows tablet

SD021-PCR-LE/WE● Distributor/Representative

1-1-3, Higashiyamata, Tsuzuki-ku, Yokohama, Kanagawa, 224-0023, Japan
Phone:(+81)45-593-0200, Facsimile:(+81)45-593-7591, https://global.kikusui.co.jp/■ All products contained in this catalogue are equipment and devices that are premised on use under the supervision of qualified
personnel, and are not designed or produced for home-use or use by general consumers. ■ Specifications, design and so forth
are subject to change without prior notice to improve the quality. ■ Product names and prices are subject to change and production
may be discontinued when necessary. ■ Product names, company names and brand names contained in this catalogue represent
the respective registered trade name or trade mark. ■ Colors, textures and so forth of photographs shown in this catalogue may
differ from actual products due to a limited fidelity in printing. ■ Although every effort has been made to provide the information as
accurate as possible for this catalogue, certain details have unavoidably been omitted due to limitations in space. ■ If you find any
misprints or errors in this catalogue, it would be appreciated if you would inform us. ■ Please contact our distributors to confirm
specifications, price, accessories or anything that may be unclear when placing an order or concluding a purchasing agreement.

Issue:Nov.2022 202211PDFEC81