Film Degassing Chamber
IDEX PN: 9000-2071
Product Datasheet

Description
The 9000-2071 is a 500 µL degassing chamber in a flat film format with ¼-28 flat bottom liquid ports.
The fluoropolymer membrane & PEEK chamber is designed to degas solvents at the highest
performance levels available for analytical scale HPLC & IC systems. In a small form factor, withreinforced Teflon™ AF film; this degassing chamber has a metal free flow path and universal solvent
compatibility, including HFIP and Hexane. The patented chamber design enables low flow restriction
when compared to the lumen based chambers.

Degassing Chamber Technical Data
Degassing Channel Membrane
Wetted Materials
Vacuum Housing Material
Vacuum Side Materials
Degassing Channel Volume
Vacuum Volume
Maximum Pressure Rating
Mounting
Liquid Connections
Vacuum Connections

Teflon™ AF, Carbon
Teflon™ AF, PEEK, Carbon
PEEK
PEEK, PFA
Nominal 500µl~100 µl
27 PSIG
Threaded 6-32 UNC-2B
Threaded ¼-28 UNF-2B flat bottom, 3/8” depth
Designed for 1/8” ID elastomeric tubing

Operating Temperature
Storage/Shipping Temperature
Maximum Flow Rate

1°C to 60°C (IPA/H2O 20/80)-30°C to 70°C
10mL/minPage 1 of 52322783 B
12/12/2019

9000-2071- Film Degassing Chamber
Dimensional Drawing
Dimensions are in Inches (millimeters)

Page 2 of 62322783 B
12/12/2019

9000-2071- Film Degassing ChamberFilm Chamber Degassing Efficiency

Page 3 of 62322783 B
12/12/2019

9000-2071- Film Degassing ChamberFilm Chamber Fluidic Resistance

Page 4 of 62322783 B
12/12/2019

9000-2071- Film Degassing Chamber
Chemical Compatibility
CHEMICAL COMPATIBILTY
DESCRIPTIONDEGASSER COMPATIBILITY
Acetic Acid, Glacial
Compatible
AcetoneCompatible
Acetonitrile
Compatible
Chlorobenzene
Compatible
Chloroform
Compatible
Dichloromethane
Compatible
Dimethyl Sulfoxide
Compatible
Ether
Compatible
Ethyl Acetate
Compatible
Hexafluoroisopropanol
Compatible 3
Hexanes (60% n-Hexane)
Compatible
Isopropyl Alcohol
Compatible
Methyl Alcohol, Anhydrous
Compatible
Methyl Ethyl Ketone
Compatible
Methyl Isobutyl Ketone
Compatible
Methyl tert-Butyl Ether
Compatible
N,N-Dimethylacetamide
Compatible
N,N-Dimethylformamide
Compatible
n-Propyl Alcohol
Compatible
NMP (N-Methyl-2-pyrrolidinone)
Compatible
Sodium Azide
Incompatible
TetrahydrofuranCompatible
TolueneCompatible
Water
Compatible

Page 5 of 62322783 B
12/12/2019

9000-2071- Film Degassing Chamber
Buffers

GENERAL CHEMICALS COMPATIBILTY
Compatible 2

Hydro fluoro solvents
Perfluorinated solvents
FreonsSalts
Acids
Bases

Incompatible
Incompatible
Incompatible
Compatible 2
Compatible 1
Compatible 1

Notes:

1. Concentrated acids and bases, particularly 1 molar or greater may attack PEEK and PPS wetted parts.
Exposure to these acids should be avoided.
2. Highly concentrate buffer and salt containing mobile phases should be washed from the HPLC flow pathincluding the degasser flow path if the instrument is to be idle for an extended period. Particular attention
should be paid to not leave such concentrated solutions in an unused flow path for an extended period.
Crystallization may occur resulting in blockage of the flow path.
3. HFIP compatibility is one channel pure HFIP and any number of channels are compatible with HFIP inwater.
The above chemical compatibility information has been obtained from our material suppliers, third party
sources, and internal evaluations.
IDEX Health & Science LLC has not performed chemical compatibility testing on all solvents listed.

Regulatory Compliance Information
This product is compliant with current RoHS & REACH regulations

Country of Origin
Manufactured & Assembled in USA

Reference Document
Constant Performance HPLC Degassing White Paper
Film Degasser Brochure
12/12/2019
2322783 B

Page 6 of 6