CERTIFICATE(1)

EU-Type Examination(2)

Equipment or protective systems intended for use inpotentially explosive atmospheres - Directive
2014/34/EU(3)

EU-Type Examination Certificate Number: DEKRA 18ATEX0130(4)

Product:

Portable Gas Monitor Model GW-3
(CMS-4 when marked for the Clemco Industries Corp.)(5)

Manufacturer:

RIKEN KEIKI Co., Ltd.

(6)

Address:

2-7-6, Azusawa, Itabashi-ku, Tokyo, 174-8744, Japan(7)

This product and any acceptable variation thereto is specified in the schedule to this certificate and the documents
therein referred to.

(8)

DEKRA Certification B.V., Notified Body number 0344 in accordance with Article 17 of Directive 2014/34/EU of theEuropean Parliament and of the Council, dated 26 February 2014, certifies that this product has been found tocomply with the Essential Health and Safety Requirements relating to the design and construction of products
intended for use in potentially explosive atmospheres given in Annex II to the Directive.

Issue Number: 3

The examination and test results are recorded in confidential test report number NL/DEK/ExTR18.0075/03.
(9)Compliance with the Essential Health and Safety Requirements has been assured by compliance with:EN IEC 60079-0 : 2018

EN 60079-11 : 2012

except in respect of those requirements listed at item 18 of the Schedule.
(10)

If the sign ”X” is placed after the certificate number, it indicates that the product is subject to the Specific Conditionsof Use specified in the schedule to this certificate.

(11)This EU-Type Examination Certificate relates only to the design and construction of the specified product. Further
requirements of the Directive apply to the manufacturing process and supply of this product. These are not coveredby this certificate.

(12)

The marking of the product shall include the following:

II 1 GEx ia IIC T4 Ga

Date of certification: 24 April 2024
DEKRA Certification B.V.

ÙL.G. van Schie
Certification ManagerThroughout this document, a point is used as the decimal separator.
© Integral publication of this certificate and adjoining reports is allowed. This Certificate may only be reproduced
in its entirety and without any change.

DEKRA Certification B.V. Meander 1051, 6825 MJ Arnhem P.O. Box 5185, 6802 ED Arnhem The Netherlands
T +31 88 96 83000 F +31 88 96 83100 www.dekra-product-safety.com Registered Arnhem 09085396

Page 1/2

(13)SCHEDULE(14)

to EU-Type Examination Certificate DEKRA 18ATEX0130(15)

Description

Issue No. 3

The Portable Gas Monitor model GW-3 (CMS-4 when marked for the Clemco Industries Corp.) is
designed to provide continuous exposure monitoring of toxic gases and/or oxygen by diffusion
sampling method.
Portable Gas Monitor is available either as a single gas monitor for measuring carbon monoxide(CO), hydrogen sulphide (H2S) or oxygen (O2), or as dual gas monitor for measuring
simoultaneosly (CO/O2) or (CO/H2).
Sensor types used are either electrochemical or galvanic type.
The measurement results are displayed on the LCD while gas alarms can be issued via LED andbuzzer. Measurement results can be wirelessly transmitted to a host using IrDA.
The examination of the Portable Gas Monitor does not include a judgement of the functional
performance of the apparatus.
Ambient temperature: -20 °C to +60 °C.
Electrical data
Supply:User-replaceable primary lithium manganese button cell battery (3.7 V, 610 mAh).
Installation instructions
The instructions provided with the product shall be followed in detail to assure safe operation.
(16)

Report Number
NL/DEK/ExTR18.0075/03.

(17)

Specific conditions of use
None.

(18)

Essential Health and Safety Requirements
Covered by the standards listed at item (9).

(19)

Test documentation
As listed in Report No. NL/DEK/ExTR18.0075/03.

AM 2972:4

Page 2/2