SPD1000XSeries

DataSheet-2021.04

Programmable DCPower SupplySPD1000X Series Programmable Linear DC Power Supply

Main Features
Single path high-precision programmable voltage

SPD1168X
SPD1305X

output：SPD1168X: 16 V/8 A, total power up to 128 W
SPD1305X: 30 V/5 A, total power up to 150 W
Stable, reliable, Low ripple and noise：≤ 350 uVrms/3mVpp; < 2 mArms
Fast transient response time: < 50 μs

Product Overview
SPD1000X Programmable Linear DC Power Supply hasa 2.8 inch TFT-LCD display, features remote computer
control capability, and real time wave display, to deliver
high performance and ease-of-use.
The SPD1168X features a high precision programmable
output capable of delivering up to 16 V, the SPD1305X
features a high precision programmable output capable

5 digit Voltage, 4 digit Current Display, Minimum
Resolution: 1 mV/1 mA
Supports front panel timing output functions
2.8 inch true color TFT- LCD 240 *320 display
2 types of output modes：Two-wire output mode, 4-wirecompensation output mode，Maximum compensation
voltage 1V.
100/120/220/230 V compatible design to meet the needsof delivering up to 30 V and also includes a 4-wire senseof different power grids

function for more accurate voltage sourcing, especiallyIntelligent temperature-controlled fanreduces noisefor long leads or high resistance connections. There are
additional output short and overload protect functions toassist in production and development applications.

Clear graphical interface, with the waveform display
functionInternal 5 groups of system parameter save/recall
Includes PC software: Easypower, supports SCPI, LabView
driverWWW.SIGLENT.COM

SPD1000X Series Programmable Linear DC Power Supply

Design Features
High-resolution and high-precision output
The SPD1000X power supply features a high measurement resolution of 1mV/1mA . This ensures accurate output even with very with small changes in voltage orcurrent. This is impossible for a low resolution power supply.

4-wire SENSE compensation mode function
In the 4-wire SENSE compensation output mode: By using a separate measurement circuit, the supply can more accurately compensate for any voltage drops due tohigh resistance connections or long cables. Maximum compensation voltage is 1V.

Low ripple and noise

Fast transient response time

Low voltage overshoot

0.01% Load Regulation & 0.2% LineRegulation

Panel displays the timing output

Save/Recall setting parameters
SPD1000X programmable power supply can save or recall 5groups of setting parameters in internal storage. You can easily
recall the settings you need.

Panel timing output

Real time wave
displayInternal Stora

PC Timer
WWW.SIGLENT.COM

SPD1000X Series Programmable Linear DC Power SupplySpecifications
All the specifications are guaranteed when the instrument has been working for more than 30 minutes under the specified operating temperature. Unless otherwise
noted, the specifications are applicable to all the channels of the specified model.

ModelSPD1168X

SPD1305X

DC Output(0 ℃ to 40 ℃ )Output Voltage: 0 to 16 V
Output Current: 0 to 8 A

Output Voltage: 0 to 30 V
Output Current: 0 to 5 A

Max Output Power

128W

150W

Display

2.8 inch true color TFT-LCD
5 digit voltage/4 digit current

Resolution

1mV/1mA
Voltage: ±(0.03% of reading+10 mV)

Program Accuracy(25 ± 5 ℃ )Current: ±(0. 3% of reading+10 mA)
Voltage: ±(0.03% of reading+10 mV)

Readback Accuracy(25 ± 5 ℃ )Current: ±(0. 3% of reading+10 mA)

Temperature Coefficient per ℃ (Output Percentage+ Offset)

Constant Voltage
ModeConstant Current
ModeVoltage: ±(0.01% of reading+3 mV)Current: ±(0.01% of reading+3 mA)

Load Regulation≤ 0.01% + 2 mV

Ripple & Noise≤ 350 uVrms/3 mVpp (20 Hz to 20 MHz)

Recovery Time

< 50 μs (50% load change, minimum load 0.5 A)

Line Regulation≤ 0.2% + 3 mA

Load Regulation≤ 0.2% + 3 mA

Ripple & Noise≤ 2 mArms

Locking Key

Yes

Memory Save/Recall5 Sets

Power Source

AC 100 /120/220/230 V ± 10% 50/60Hz

Standard Configuration Interface

USB Device, LANInsulation

Case to Terminal ≥ 20 MΩ (DC 500 V)
Case to AC line ≥ 30 MΩ (DC 500 V)

Operating Environment

Outdoor Usage:
Elevation: ≤2000 m Environment Temperature 0 to 40 ℃ Relative Humidity ≤ 80%
Installation Level: Ⅱ Pollution Level: 2Storage Environment

Environment Temperature: -10 to 70 ℃ Relative Humidity ≤ 70%

Dimension

154.6 (W) × 144.5 (H) × 280(D) mm

Weight≈5.5kgOrdering information
Product information

Single path independent output, min resolution 1 mV/1 mA, USB Device & LAN, 2.8 inch LCD display

Standard Accessories
USB Cable -1

Quick Start -1
Power cord -1
Output Test Cord -2 Sets
Certificate of Calibration -1

Warranty
Three-year warranty, excluding accessories.

WWW.SIGLENT.COM

Product No

SPD1168X, SPD1305X

SPD1000X
SeriesProgrammable DC
Power Supply

About SIGLENTSIGLENT is an international high-tech company, concentrating on R&D, sales,production and services of electronic test & measurement instruments.

SIGLENT ﬁrst began developing digital oscilloscopes independently in 2002.
After more than a decade of continuous development, SIGLENT has extended
its product line to include digital oscilloscopes, function/arbitrary waveformgenerators, RF generators, digital multimeters, DC power supplies, spectrum
analyzers, vector network analyzers, isolated handheld oscilloscopes,
electronic load and other general purpose test instrumentation. Since its ﬁrst
oscilloscope, the ADS7000 series, was launched in 2005, SIGLENT has become
the fastest growing manufacturer of digital oscilloscopes. We ﬁrmly believethat today SIGLENT is the best value in electronic test & measurement.

Headquarters:
SIGLENT Technologies Co., LtdAdd: Bldg No.4 & No.5, Antongda IndustrialZone, 3rd Liuxian Road, Bao'an District,Shenzhen, 518101, China
Tel: + 86 755 3688 7876
Fax: + 86 755 3359 1582
Email: sales@siglent.com
Website: int.siglent.com
USA:
SIGLENT Technologies America, Inc6557 Cochran Rd Solon, Ohio 44139
Tel: 440-398-5800
Toll Free: 877-515-5551
Fax: 440-399-1211
Email: info@siglent.com
Website: www.siglentna.com

Europe:SIGLENT Technologies Germany GmbH
Add: Staetzlinger Str. 7086165 Augsburg, Germany
Tel: +49(0)-821-666 0 111 0Fax: +49(0)-821-666 0 111 22
Email: info-eu@siglent.com
Website: www.siglenteu.com

Follow us on
Facebook: SiglentTech