SYN6464-channel
synchronization
distributor
Data SheetEN01B

SYN64

Key Features
1:64 high-speed fanout
DC-coupled and AC-coupled input modes-10V ~ 10V input range-10V ~ 10V trigger level range
Low jitter, low skew
1U height for standard rack mounting

Product Overview
The SYN64 synchronization distributor is a
1:64 high-speed digital signal driver, which
fan-outs the input into 64 outputs with low-jitter,
low-skew. Its input offers both DC and AC
coupling modes and adjustable trigger level.
For a multi-channel acquisition system builtwith multiple oscilloscopes, a synchronization
distributor is an essential device. Use theSYN64 as the trigger source of theoscilloscopes, connecting the outputs of theSYN64 to the EXT trigger or analog channelof the oscilloscopes to build an acquisition
system. You can also use the SYN64 tocalibrate the skew between channels of theacquisition system. By connecting the outputsof the SYN64 to the analog channel inputs of
the oscilloscopes, you can calibrate the delayof each analog channel and cable to ensure
the alignment of multiple channels.

SYN64 Data Sheet

1Application
Multi-channel High-speed Acquisition SystemConnection
DiagramSync

HostLAN connectiion
Synchronization signal
10MHz In (optional)

SYN64

Oscilloscope #1

Oscilloscope #2

Network Switch

Oscilloscope #3

10MHz Out (optional)

Oscilloscope #N

Multiple units are combined to create a high-speed acquisition system with up to 512 channels by
being triggered with low-skew synchronization signals from the SYN64

The host can access each unit over 1000M LAN. A complete SCPI command set as well as
LabVIEW and IVI drivers are provided for easy data acquisition. The LAN port is LXI compliant.

Sample clocks are synchronized between all units in the test system by cascading the 10 MHz In
and 10 MHz Out clocks in a daisy chain

SYN64 Data Sheet2

Mechanical Dimensions

SYN64 Data Sheet

3

Specifications
All specifications are not guaranteed unless the following conditions are met:
The module has been working continuously for at least 30 minutes at the specified temperature (18℃~ 28℃)

Input
Channel1

Voltage range-10 V ~ +10 VSwing

1.5 ~ 20 Vpp

Impedance

50 ΩPowerMinimum pulsewidthCoupling≤ 1W

Trigger level-10 V ~ +10 V, adjusted by potentiometer

Connector

SMA

1 nsDC, ACOutput
Channel64Group

8. Output 1~8 as Group 1, Output 9~16 as Group 2, and so on

Voltage

800 mVpp swing, 400 mV DC offset (into 50 Ω)

Rise time

300 ps typical (20% to 80%)

Jitter1 ps rms typical

Skew

In group: 30 ps typical, 50 ps maximum, 0 ~ 50 ℃
Between groups: 140 ps typical, 200 ps maximum, 0 ~ 50 ℃

Requirement onload
Connector

50 Ω to ground, DC coupled
SMA

GeneralDimensions
Powerconsumption
Temperature
Humidity
Altitude

SYN64 Data Sheet

Length × Height × Width =434.8 mm × 43.0 mm × 285.4 mm
12 W
Operating: 0℃ ~ 50℃
Non-operating: -30℃ ~ 70℃
Operating: 5% ~ 90%RH, 30℃, degraded to 50%RH at 40℃
Non-operating: 5% ~ 95%
Operating: ≤ 3,048 m, 25℃
Non-operating: ≤15,266 m

4

About SIGLENTSIGLENT is an international high-tech company, concentrating on R&D, sales,
production and services of electronic test & measurement instruments.

SIGLENT first began developing digital oscilloscopes independently in 2002.
After more than a decade of continuous development, SIGLENT has extended

its product line to include digital oscilloscopes, isolated handheld
oscilloscopes,

function/arbitrary

waveform

generators,

RF/MWsignalgenerators, spectrum analyzers, vector network analyzers, digital multimeters,
DC power supplies, electronic loads and other general purpose test
instrumentation. Since its first oscilloscope was launched in 2005, SIGLENT
has become the fastest growing manufacturer of digital oscilloscopes. We
firmly believe that today SIGLENT is the best value in electronic test &measurement.

Headquarters:
SIGLENT Technologies Co., Ltd
Add: Bldg No.4 & No.5, Antongda IndustrialZone, 3rd Liuxian Road, Bao'an District,Shenzhen, 518101, China
Tel: + 86 755 3688 7876
Fax: + 86 755 3359 1582
Email: sales@siglent.com
Website: int.siglent.com

North America:
SIGLENT Technologies America, Inc6557 Cochran Rd Solon, Ohio 44139
Tel: 440-398-5800
Toll Free: 877-515-5551
Fax: 440-399-1211
Email: info@siglentna.com
Website: www.siglentna.com

Follow us on
Facebook: SiglentTech

Europe:SIGLENT Technologies Germany GmbH
Add: Staetzlinger Str. 70
86165 Augsburg, Germany
Tel: +49(0)-821-666 0 111 0
Fax: +49(0)-821-666 0 111 22
Email: info-eu@siglent.com
Website: www.siglenteu.com

SYN64 Data Sheet5