SDS2000X-E
Digital Oscilloscope

Data Sheet
Rev. 02B

Key Features
200 MHz, 350 MHz bandwidth models
Real-time sampling rate up to 2 GSa/s
The Siglent SPO technology

SDS2202X-E
SDS2352X-E◼

Waveform capture rates up to 110,000 wfm/s (normal mode)and 400,000 wfm/s (sequence mode)◼

Supports 256-level intensity grading and color temperature
display modes◼

Record length up to 28 Mpts◼

Digital trigger system

Intelligent trigger: Edge, Slope, Pulse Width, Window, Runt,
Interval, Time out (Dropout), Pattern
Serial bus triggering and decoding (Standard), supports protocols

Product Overview
SIGLENT’s SDS2000X-E Series Super Phosphor Oscilloscopes are available inthree bandwidths, 100 MHz, 200 MHz and 350 MHz, have a sampling rate of
2 GSa/s and a standard record length of 28 Mpts. The most commonly usedfunctions can be accessed with its user-friendly one-button design.
The SDS2000X-E series employs a new generation of SPO technology. With its
excellent signal fidelity, background noise is lower than similar products in
the industry. It has a minimum vertical input range of 500 uV/div, aninnovative digital trigger system with high sensitivity and low jitter, and a
waveform capture rate of 400,000 frames/sec (sequence mode). It also
employs not only the common 256-level intensity grading display function
but also a color temperature display mode not found in other models in thisclass. Siglent’s new oscilloscopes offering supports multiple powerfultriggering modes including serial bus triggering and decoding. History
waveform recording and sequential triggering allow for extended waveformrecords to be captured, stored, and analyzed. With 1M points of FFT,SDS2000X-E has very high spectrum resolution. With hardware coprocessor,
it executes true fast measurement and math to all of 28M sample points sothat there is no any distortion on analysis. It also supports searching andnavigating, on-screen Bode plot, 16 digital channels(Option), an external
USB powered 25 MHz AWG module(Option), a USB WIFI adapter(Option), andan embedded application that allows remote control via web browser. The
features and high-performance of the SDS2000X-E oscilloscopes cannot bematched else anywhere at this price.

IIC, SPI, UART, CAN, LIN
Video trigger, supports HDTV
Low background noise with voltage scales from 500 μV/div to 10
V/div
10 types of one-button shortcuts, supports Auto Setup, Default,
Cursors, Measure, Roll, History, Display/Persist, Clear Sweep, Zoom
and Print
Segmented acquisition (Sequence) mode, divides the maximum
record length into multiple segments (up to 80,000), according totrigger conditions set by the user, with a very small dead time
segment to capture the qualifying event
History waveform record (History) function (maximum recorded
waveform length is 80,000 frames)
Automatic measurement function for 38 parameters as well as
Measurement Statistics, Zoom, Gating, Math, History and Referencefunctions
1 Mpts FFT with Peaks and Markers
Math and measurement functions use all sampled data points (upto 28 Mpts)
Math functions (FFT, addition, subtraction, multiplication, division,
integration, differential, square root)
Preset key can be customized for user settings or factory“defaults”
Security Erase mode
High Speed hardware-based Pass/ Fail function
MSO, 16 digital channels (option)
Bode plot, Measuring Power Supply Control Loop Response
Search and navigate
USB AWG module (option)
USB WIFI adapter (option)
Web Browser based control
Large 7-inch TFT-LCD display with 800 * 480 resolution
Multiple interface types: USB Host, USB Device (USB -TMC), LAN,
Pass / Fail, Trigger Out
Supports SCPI remote control commands
VXI-11+SCPI, Telnet (Port 5024) +SCPI and Socket (Port 5025) +SCPI
programming over LAN
Supports web control and virtual instrument control panel for both
PC and mobile terminals
Supports Multi-language display and embedded online help

WWW.SIGLENT.COM
1

SDS2000X-E Series Digital Oscilloscope

Models and Key Specifications
Model

SDS2202X-E

Bandwidth

200 MHz

Sample rate (Max.)

2 GSa/s

Channels

2+EXT

Memory depth (Max.)

SDS2352X-E
350 MHz

14Mpts/CH(not interleave mode);
28Mpts/CH(interleave mode)

Waveform capture rate (Max.)

110,000 wfm/s (normal mode), 400,000 wfm/s (sequence mode)

Trigger type

Edge, Slope, Pulse Width, Window, Runt, Interval, Dropout, Pattern, Video

Serial Trigger and decoder (Std)

IIC, SPI, UART, CAN, LIN

16 Digital Channels (option)

Maximum waveform capture rate up to 1 GSa/s, Record length up to 14 Mpts/CH

USB AWG module (option)
Bode plot

One channel, 25 MHz, sample rate of 125 MHz, wave length of 16 kpts,
isolated output (SAG1021I only)
Minimum start frequency of 10 Hz, minimum scan bandwith of 500 Hz, maximum scan bandwidth of 120
MHz (dependent on Oscilloscope and AWG bandwidth), 500 maximum scan frequency points

USB WIFI adapter (option)

802.11b/g/n, WPA-PSK, the adapter must be supplied by Siglent to ensure working

I/O

USB Host, USB Device, LAN, Pass/Fail, Trigger Out, Sbus (Siglent MSO)

Probe (Std)2 pcs passive probe PP215

Display

7-inch TFT-LCD (800x480)

WeightWithout package 2.6Kg; With package 3.8Kg

2 pcs passive probe SP2035

Functions & Characteristics
7 Inch TFT-LCD Display and 10 One-button Menus

Front panel
7-inch TFT -LCD display with 800 * 480 resolution. Most commonly used functions are accessible using 10 different one-button operation keys:
Auto Setup, Default, Cursor, Measure, Roll, History, Persist, Clear Sweep, Zoom, Print.

2

WWW.SIGLENT.COM

Maximum sample rate of 2 GSa/s ，record Length of

Waveform Capture Rate up to 400,000 wfm/s

up to 28Mpts

Using hardware-based Zoom technologies and max record length ofWith a waveform capture rate of up to 400,000 wfm/s (sequence

up to 28 Mpts, users are able to oversample to capture for longer timemode), the oscilloscope can easily capture the unusual or low-

periods at higher resolution and use the zoom feature to see moreprobability events.

details within each signal.

Serial Bus Decoding Function (Standard)

History Waveforms (History) Mode and Segmented
Acquisition (Sequence)

SDS2000X-E displays the decoding through the events list. Bus protocol

Playback the latest triggered events using the history function.

information can be quickly and intuitively displayed in a tabular format.

Segmented memory collection will store trigger events into multiple(Up to 80,000) memory segments, each segment will store triggered
waveforms and timestamp of each frame.

True measurement to 28 M points

SDS2000X-E can measure all sampled data points up to 28 Mpts. ThisThe new math co-processor enables FFT analysis of incoming signals

ensures the accuracy of measurements while the math co-processor

using up to 1 M samples per waveform. This provides high frequency

WWW.SIGLENT.COM
3

1M points used to calculate the FFT

SDS2000X-E Series Digital Oscilloscope
decreases measurement time and increases ease-of-use.

resolution with a fast refresh rate. The FFT function also supports avariety of window functions so that it can adapt to different spectrum
measurement needs. It supports Peaks, Markers, a variety of numbers.

Hardware-Based High-Speed Pass/Fail

16 Digital Channels/MSO (option)

The SDS2000X-E utilizes a hardware-based Pass/Fail function,

16 digital channels enables users to acquire and trigger on theperforming up to 40,000 Pass / Fail decisions each second. Easily

waveforms then analyze the pattern, simultaneously with one

generate user defined test templates provide trace mask comparison

instrument.

making it suitable for long-term signal monitoring or automatedproduction line testing.

Search and Navigate

USB 25MHz AWG Module (option)

The SDS2000X-E can search events specified by the user in a frame. It

The SDS2000X-E supports a USB 25 MHz function/arbitrary waveform

can also navigate by time (delay position) and historical frames.

generator that is operated from the USB host connection. Functions
include Sine, Square, Ramp, Pulse, Noise, DC and 45 built-in
waveforms. The arbitrary waveforms can be accessed and edited by the
SIGLENT EasyWave PC software.

Bode PlotSDS2000X-E can control the USB AWG module or control an
independent SIGLENT SDG instrument, scan a devices amplitude andphase frequency response, and display the data as a Bode Plot. Thereis also a Vari-level Mode for accurately measuring Power Supply
Control Loop Response (PSRR). It can also show the result lists, andexport the data to a USB disk.

4

WWW.SIGLENT.COM

USB WIFI Adapter (option)

Web control

WiFi control of instrumentation can provide a convenient and safeWith the new embedded web server, users can control the SDS2000X-E

method of configuring and collecting data. This new feature works withfroma SIGLENT approved WiFi adapter to provide wireless control andtroubleshooting and monitoring capabilities. The web page has PC andcommunications with SIGLENT SDS2000X-E scopes. The adapter must

mobile styles that include an embedded virtual control panel.

asimpleweb

be supplied by Siglent to ensure working.

Complete Connectivity

Back panel
SDS2000X-E supports USB Host, USB Device (USB -TMC), LAN (VXI -11), Pass/Fail and Trigger Out.

Specifications
Acquisition System
Sampling Rate (Max.)

2GSa/s(single channel), 1GSa/s(two channels)

Memory Depth (Max.)Max 28Mpts/Ch(single channel), 7Mpts/Ch(two channels)

Peak Detect

1 ns

Average

Averages:4,16, 32,64,128,256,512,1024

ERES

Enhance bits:0.5, 1.5, 2, 2.5, 3

Waveform interpolationSin(x)/x, Linear

Input
Channels

2+EXT

Coupling

DC, AC, GND

WWW.SIGLENT.COM
5

page.

This

provideswonderfulremoteSDS2000X-E Series Digital Oscilloscope
Impedance
Max. Input voltage

DC1MΩ: (1MΩ±2%) || (18pF ±2pF)
DC50Ω: 50Ω±2%1MΩ: ≤400Vpk(DC + Peak AC <=10kHz)
50Ω: ≤5Vrms

CH to CH Isolation

DC-Max BW :>40 dB

Probe attenuation

0.1X,0.2X,0.5X,1X,2X,5X,10X……1000X,2000X,5000X,10000X,Custom

Vertical System
Bandwidth (-3dB)
Vertical Resolution
Vertical Scale (Probe 1X)

350MHz (SDS2352X-E)
200MHz (SDS2202X-E)
8-bit
500 μV/div–10 V/div (1-2-5 sequence )500 μV- 118 mV: ± 2 V
500μV- 100mV: ± 2V

Offset Range (Probe 1X)

102mV- 1V: ± 20V
1.02V- 10V: ± 200V

Bandwidth limit

20 MHz ± 40%

Channel Flatness

DC- 60%(BW): ± 1dB(Inner 50Ω)

60%- 100%(BW): + 1dB/-3dB

Low frequency response(AC coupling -3 dB)≤2 Hz (at input BNC)
ST-DEV ≤0.5 division (<1 mV/div)

Noise

ST-DEV ≤0.2 division (<2 mV/div)
ST-DEV ≤0.1 division(≥ 2 mV/div)

SFDR including harmonics
DC Gain Accuracy
Offset Accuracy
Rise timeOvershoot (500 ps Pulse)≥ 35 dB≤±3.0%: 5 mV/div-10 V/div≤±4.0%: ≤2 mV/div±(1%* Offset+1.5%*8*div+2 mV): ≥2 mV/div±(1%* Offset+1.5%*8*div+500 uV): ≤1 mv/div
Typical 1.0ns(SDS2352X-E)
Typical 1.8ns(SDS2202X-E)< 10%

Horizontal System
Timebase Scale

500ps/div-100s/div

Channel Skew

<100 psWaveform Capture Rate

Up to 110,000 wfm/s (normal mode), 400,000 wfm/s (sequence mode)

Intensity grading

256 Levels

Display Format

Y -T, X -Y, Roll

Timebase Accuracy±25 ppm

Roll Mode

50 ms/div-100 s/div (1-2-5 sequence)

Trigger System
Mode

Auto, Normal, Single
Internal: ±4.5 div from the center of the screen

Level

EXT: ±0.6 V
EXT/5: ±3V

Hold off range

80 ns- 1.5 s
ACCoupling

DC
LFRJ
HFRJ

6

WWW.SIGLENT.COM

Noise RJ
DC: Passes all components of the signal
Coupling

Frequency

Response

AC: Blocks DC components and attenuates signals below 8Hz
LFRJ: Blocks the DC component and attenuates the low-frequency
components below 2 MHz
HFRJ: Attenuates the high-frequency components above 1.2 MHz
DC: Passes all components of the signal

Coupling

Frequency

Response

AC: Blocks DC components and attenuates signals below 10Hz
LFRJ: Blocks the DC component and attenuates the low-frequency
components below 6 KHz
HFRJ: Attenuates the high-frequency components above 200 KHz

Accuracy (typical)

Internal: ±0.2 div
EXT: ±0.4 div
DC - Max BW 0.6div
EXT: 200mVpp DC– 10MHz

Sensitivity

300mVpp 10MHz - BW frequency(External 50Ω)
EXT/5: 1Vpp DC – 10MHz；
1.5Vpp 10MHz -BW frequency(External 50Ω)Jitter
Displacement<100 ps
Pre-Trigger: 0 - 100% Memory
Delay Trigger: 0 to 10,000 div

Edge Trigger
Slope

Rising, Falling, Rising & Falling

SourceAll channels/EXT/(EXT/5)/AC Line

Slope Trigger
Slope

Rising, Falling

Limit Range＜, ＞, ＜＞, ＞＜

SourceAll channels

Time Range

2ns- 4.2s

Resolution

1ns

Pulse Width Trigger
Polarity+wid , -wid

Limit Range＜, ＞, ＜＞, ＞＜

SourceAll channels

Pulse Range

2 ns - 4.2s

Resolution

1 ns

Video Trigger
Signal Standard

NTSC, PAL,720p/50,720p/60,1080p/50,1080p/60,1080i/50,1080i/60,Custom

SourceAll channels

Sync

Any, SelectTrigger condition

Line, Field

Window Trigger
Window Type

Absolute, Relative

SourceAll channels

Interval Trigger
Slope

Rising, Falling

Limit Range＜, ＞, ＜＞, ＞＜

SourceAll channels

Time Range

2 ns - 4.2 s

Resolution

1 ns

WWW.SIGLENT.COM
7

SDS2000X-E Series Digital Oscilloscope

Dropout Trigger
Timeout Type

Edge, State

SourceAll channels

Slope

Rising, Falling

Time Range

2 ns - 4.2 s

Resolution

1ns

Runt Trigger
Polarity+wid , -wid

Limit Range＜, ＞, ＜＞, ＞＜

SourceAll channels

Time Range

2 ns - 4.2 s

Resolution

1 ns

Pattern Trigger
Pattern Setting

Invalid, Low, High

Logic

AND, OR, NAND, NOR

SourceAll channels

Limit Range＜, ＞, ＜＞, ＞＜Time Range

2 ns - 4.2 s

Resolution

1 ns

Serial Trigger
I2C Trigger
Condition

Start, Stop, Restart, No Ack, EEPROM, 7-bits Address & Data, 10-bits Address & Data, Data Length

Source(SDA/SCL)

All channels

Data format

Hex

Limit Range

EEPROM: =, >, <Data Length

EEPROM: 1byte
Addr & Data: 1-2byte
Data Length: 1-12byte

R/W bit

Addr & Data: Read, Write, Do not care

SPI Trigger
Condition

Data

Source(CS/CL/Data)

All channels

Data format

BinaryData Length

4-96-bit

Bit Value0, 1, X

Bit Order

LSB, MSBUART Trigger
Condition

Start, Stop, Data, Parity Error

Source(RX/TX)

All channels

Data format

Hex

Limit Range=, >, <

Data Length

1 byteData Width5, 6, 7, 8-bits

Parity Check

None, Odd, Even, Space, Mark

Stop Bit

1, 1.5, 2-bits

Idle Level

High, Low

Baud Rate(Selectable)

600/1200/2400/4800/960019200/38400/57600/115200/Custom bit/s

Baud Rate (Custom)300-5000000 bit/s

CAN Trigger

8

Condition

Start, Remote, ID, ID + Data, Error

SourceAll channels

WWW.SIGLENT.COM

ID

STD (11-bits), EXT (29-bit)

Data Format

Hex

Data Length

1 -2 byte

Baud Rate

5k/10k/20k/50k/100k/125k/250k/500k/800k/1M/Custom bit/s

LIN Trigger
Condition

Break, Frame ID, ID+Data, Error

SourceAll channels

ID

1byte

Data Format

Hex

Data Length

1-2byteBaud Rate(Selectable)
Baud Rate (Custom)

600/1200/2400/4800/9600/19200/Custom bit/s
300 bit/s -20 kbit/s

SearchEventEdge, Slope, Pulse, Interval, Runt
Y-T: 700

Event Number

ROLL: No limitation
Stop After ROLL: 700

Serial Decoder
Decoders

2

IC
2

SignalSCL, SDA

Address

7, 10 bits

Threshold-4.5 - 4.5 div

List

1- 7 lines

SPI
SignalSCL,MISO, MOSI, CS (2 channel scopes can only use 2 signal) identifiers

Edge SelectRising, Falling

Idle Level

Low, HighBit Order

MSB, LSB

Threshold-4.5 - 4.5 div

List

1- 7 lines

UART
SignalRX, TX

Data Width

5，6，7，8 bits

Parity Check

None, Odd, Even, Space, Mark

Stop Bit

1, 1.5, 2 bits

Idle Level

Low, High

Threshold-4.5 - 4.5 div

List

1- 7 lines

CAN
SignalCAN_H, CAN_L

SourceCAN_H, CAN_L, CAN_H-CAN_L

Threshold-4.5 - 4.5 div

List

1- 7 lines

LIN
LIN Specification Package

WWW.SIGLENT.COM
9

Ver1.3, Ver2.0

SDS2000X-E Series Digital Oscilloscope
Revision
Threshold-4.5 - 4.5 div

List

1- 7 lines

Measurement
SourceNumber of
Measurements

All channels, All channels in Zoom, Math, All References, History
Display 4 measurements at the same time. 5 measurements displayed in statistics table.

Measurement Range

Screen or Gate region

Measurement Parameters

38 Types
Max

Highest value in input waveform

Min

Lowest value in input waveform

Pk-Pk

Difference between maximum and minimum data values

Ampl

Difference between top and base in a bimodal signal, or between max and min in an
unimodal signal

VerticalTop

Value of most probable higher state in a bimodal waveform

Base

Value of most probable lower state in a bimodal waveform

Mean

Average of all data values

Cmean

Average of data values in the first cycle

Stdev

Standard deviation of all data values

Cstd

Standard deviation of all data values in the first cycle

VRMS

Root mean square of all data values

Crms

Root mean square of all data values in the first cycle

FOV

Overshoot after a falling edge;(base -min)/Amplitude

FPRE

Overshoot before a falling edge;(max -top)/Amplitude

ROV

Overshoot after a rising edge;(max -top)/Amplitude

RPRE

Overshoot before a rising edge;(base -min)/Amplitude

Level@X

the voltage value of the trigger point

PeriodTime between the middle threshold points of two consecutive, like-polarity edges

Freq

Reciprocal of period+Wid

Width measured at 50% level and positive slope-Wid

Width measured at 50% level and negative slope

Rise Time

Duration of rising edge from 10 -90%

Fall Time

Duration of falling edge from 90 -10%

Bwid

Time from the first rising edge to the last falling edge, or the first falling edge to the lastrising edge at the 50% crossing+Dut
Horizontal

Time difference between the 50% threshold of a rising edge to the 50% threshold of thenext falling edge of the pulse-DutTime difference between the 50% threshold of a falling edge to the 50% threshold of thenext rising edge of the pulse

DelayTime from the trigger to the first transition at the 50% crossing

Time@Level

Time from the trigger to each rising edge at the 50% crossing.
When Statistics is Off, it shows the time from the trigger to the last rising edge at the 50%
crossing.
When Statistics is On, it shows the Mean, Min, Max, Standard Deviation of time from thetrigger to each rising edge at the 50% crossing in multiple frames (number = Count). The
Current shows the time of current frame from the trigger to the last rising edge at the 50%
crossing.

Delay

10
WWW.SIGLENT.COM

Phase

Phase difference between two edges

FRR

Time from the first rising edge of channel A to the following first rising edge of channel B

FRF

Time from the first rising edge of channel A to the following first falling edge of channel

B
FFR

Time from the first falling edge of channel A to the following first rising edge of channel
B

FFF

Time from the first falling edge of channel A to the following first falling edge of channel
B

LRR

Time from the first rising edge of channel A to the last rising edge of channel B

LRF

Time from the first rising edge of channel A to the last falling edge of channel B

LFRTime from the first falling edge of channel A to the last rising edge of channel B

LFF

Time from the first falling edge of channel A to the last falling edge of channel B

Skew

Time of source A edge minus time of nearest source B edge

Manual : Time X1, X2, (X1 -X2), (1/ΔT)
Cursors

Voltage Y1, Y2, (Y1 -Y2)
Track: Time X1, X2, (X1 -X2)

Statistics

Current, Mean, Min, Max, Stdev, Count

Counter

Hardware 6-digit counter（channels are selectable）

Math
Operation+,- , * , / , FFT , d/dt , ∫dt , √FFT window

Rectangular, Blackman, Hanning, Hamming, Flattop

FFT display

Full Screen, Split, Exclusive

USB AWG Module (option)
Channel1Max. Output Frequency

25 MHzSampling Rate

125 MSa/s

Frequency Resolution

1 μHz

Frequency Accuracy±50 ppm

Vertical Resolution

14-bits

Amplitude Range-1.5 ～ +1.5 V (50Ω load)-3 ～ +3 V (High-Z load)

Waveform TypeSine, Square, Ramp, Pulse, Noise, DC and 45 built-in waveforms

Output impedance

50Ω ± 2%

Protection

Over-Voltage Protection, Current-Limiting Protection

Insulation Voltage±42 Vpk (for SAG2021I only)

Sine
Frequency

1 μHz～ 25 MHz

Offset Accuracy(10 kHz)±(1%*Offset Setting Value +3 mVpp)

Amplitude flatness±0.3 dB(10 kHz, 5Vpp)
SFDR

DC～1 MHz-60dBc1 MHz～5 MHz-55dBc5 MHz～25 MHz -50dBc
HD

DC～5 MHz-50dBc5 MHz～25 MHz -45dBc

Square/Pulse
Frequency

1 μHz～ 1 0MHz

Duty Cycle

1% ～99%Rise/Fall time< 24 ns (10% ～ 90%)

Overshoot(1kHz,1Vpp,< 3% (typical 1kHz, 1 Vpp)

Typical)WWW.SIGLENT.COM
11

SDS2000X-E Series Digital Oscilloscope
Pulse Width> 50 ns

Jitter< 500 ps + 10 ppm

Ram
Frequency

1 μHz～ 300 kHz

Linearity(Typical)< 0.1% of Pk-Pk (Typical, 1 kHz, 1 Vpp, 50% Symmetry)

Symmetry

0% ～ 100%

DC
Offset range±1.5 V (50Ω load)±3 V (High-Z load)

Accuracy±(|offset|*1%+3 mV)

NoiseBandwidth>25 MHz (-3dB)

Arbitrary Wave
Frequency

1 μHz～ 5 MHz

Wave Length

16 kpts

Sampling Rate

125MSa/s

Lead in

EasyWave and U-Disk

Digital Channels (option)
No. of Channels

16

Max. Sampling Rate

1 GSa/sMemory Depth

14 Mpts/CH

Min.

Detectable

Pulse

Width

4 ns

Level GroupD0~D7, D8~D15

Level Range-8 V ~ 8 V

Logic Type

TTL, CMOS, LVCMOS3.3, LVCMOS2.5, Custom

Skew

D0~D15: ±1 sampling interval
Digital to Analog: ± (1 sampling interval +1ns)

I/O
Standard

USB Host*2, USB Device, LAN, Pass/Fail, Trigger Out

Pass/Fail

3.3V TTL Output

Display(Screen)
Display Type

7–inch TFT LCD

Display Resolution

800×480 pixels

Display Color

24-bitContrast(Typical)

500:1

Backlight

300 nits

Display(Waveform)Range

8 x 14 divisions

Display Mode

Dot, Vector

Persist Time

Off, 1 Sec, 5 Sec, 10 Sec, 30 Sec, Infinite

Color Display

Normal, Color

Screen Saver

1 min, 5 min, 10 min, 30 min, 1 hour, Off

Language

Simplified Chinese, Traditional Chinese, English, French, Japanese, Korean, German, Russian, Italian, Portuguese

12
WWW.SIGLENT.COM

Environments
Temperature
Humidity

HeightOperating: 0℃ - +40℃
Non-operating: -20℃ - + 60℃
Operating: 85% RH, 40 ℃, 24 hours
Non-operating: 85% RH, 65 ℃, 24 hours
Operating: ≤ 3000 m
Non-operating: ≤ 15,000 m

Standards
Electromagnetic

Meets EMC directive (2014/30/EU), meets or exceeds IEC 61326-1:2012/EN61326-1:2013 (Basic)compatibility

Conducted disturbance

CISPR 11/EN 55011

CLASS A group 1，150kHz-30MHz

Radiated disturbance

CISPR 11/EN 55011

CLASS A group 1， 30MHz-1GHz

Electrostatic discharge (ESD)

IEC 61000-4-2/EN 61000-4-2

4.0 kV（Contact），8.0 kV（Air）

Radio-frequency

electromagnetic

field Immunity

10 V/m（80 MHz to 1 GHz）;
IEC 61000-4-3/EN 61000-4-3

1 V/m（2.0 GHz to 2.7GHz）

Electrical fast transients (EFT)IEC 61000-4-4/EN 61000-4-4

SurgesIEC 61000-4-5/EN 61000-4-5

Radio-frequency

3 V/m（1.4 GHz to 2 GHz）;

continuous

conducted Immunity

IEC 61000-4-6/EN 61000-4-6

2kV （Input AC Power Ports）
1kV （Line to line）
2kV （Line to ground）
3 V, 0.15-80MHz
Voltage Dips：
0% UT during 1 cycle;

Voltage dips and interruptions

IEC 61000-4-11/EN 61000-4-11

40% UT during 10/12 cycles;
70% UT during 25/30 cycles
Voltage interruptions：0% UT during
250/300 cycles

SafetyUL 61010-1:2012/R: 2018-11; CAN/CSA-C22.2 No. 61010-1:2012/A1:2018-11.
UL 61010-2-030:2018; CAN/CSA-C22.2 No. 61010-2-030:2018.

Power Supply
Input Voltage
Power

100 ～ 240 Vrms 50/60Hz
100 ～ 120 Vrms 400Hz
50 W Max

Mechanical
Length: 312 mm
Dimensions

Width: 132.6 mm
Height: 151 mm

WeightN.W: 2.6 kg; G.W: 3.8 kg

Probes and Accessories
Probe

WWW.SIGLENT.COM
13

PictureModel

Specifications &Description

SDS2000X-E Series Digital Oscilloscope

Passive

PP510

Bandwidth: 100MHz, 1X/10X, 1M/10Mohm,300V/600V

PP215

Bandwidth: 200MHz, 1X/10X, 1M/10Mohm, 300V/600V

SP2035

Bandwidth: 350MHz, 1X/10X, 1M/10Mohm, 150V/300V

Bandwidth: 100 KHz, Max. continuous current: 20Arms Peak current:
CP402060A
Switch Ratio: 50mV/A, 5mV/A, Accuracy: 50mV/A (0.4A -10Apk)±2%,
5mV/A(1A-60Apk) ±2%, 9V battery source
Bandwidth: 1MHz, Max. continuous current: 50Arms, Peak current:

CP4050140A
Switch Ratio: 500mV/A, 50mV/A
Accuracy: 500mV/A (20mA -14ApK)±3%±20mA , 50mV/A (200mA 100ApK)±4%±200mA, 50mV/A (100A -140ApK) ±15%max, 9V battery source
Bandwidth: 150kHz, Max. continuous current: 70Arms, Peak current:

CP4070200A
Switch Ratio: 50mV/A, 5mV/A, Accuracy: 50mV/A (0.4A -10ApK)±2% ,5mV/A(1A -200ApK) ±2%, 9V battery source

Current Probe

Bandwidth: 50 MHz, Max. continuous current: 30Arms, Peak current:
CP503050A
Switch Ratio: 100mV/A, 1V/A, Accuracy: 1V/A (±1%±1mA), 100mV/A(±1%±10mA), DC12V/1.2A power adapter
Bandwidth: 100 MHz, Max. continuous current: 30Arms, Peak current:

CP5030A

50A
Switch Ratio: 100mV/A, 1V/A, Accuracy: 1V/A (±1%±1mA), 100mV/A(±1%±10mA), DC12V/1.2A power adapter
Bandwidth: 12 MHz, Max. continuous current: 150Arms, Peak current:

CP5150300A
Switch Ratio: 100mV/A, 10mV/A, Accuracy: 100mV/A (±1%±10mA),
10mV/A (±1%±100mA), DC12V/1.2A power adapter
Bandwidth: 5 MHz, Max. continuous current: 500Arms, Peak current:

CP5500750A
Switch Ratio: 100mV/A, 10mV/A, Accuracy: 100mV/A (±1%±10mA),
10mV/A(±1%±100mA), DC12V/1.2A power adapter

Differential Probe

14
WWW.SIGLENT.COM

DPB4080

Bandwidth: 50MHz, Differential Range: 800V (DC + Peak AC),
100X/200X/500X/1000X, Accuracy: ±1%, DC 9V/1A power adapter

DPB5150

Bandwidth: 70MHz, Differential Range: 1500V (DC + Peak AC),50X/500X
Accuracy: ±2%, DC 5V/1A USB adapter

DPB5150A

Bandwidth: 100MHz, Differential Range: 1500V (DC + Peak AC),
50X/500X , Accuracy: ±2%
DC 5V/1A USB adapter

DPB5700

Bandwidth: 70MHz, Differential Range: 7000V (DC + Peak AC),
100X/1000X , Accuracy: ±2%,
DC 5V/1A USB adapter
Bandwidth: 100MHz

DPB5700A

Differential Range: 7000V (DC + Peak AC),
100X/1000X
Accuracy: ±2%
DC 5V/1A USB adapter
Bandwidth: 40MHzHigh Voltage

HPB4010Differential Range: DC 10kV, AC (rms): 7kV (sine), AC (Vpp): 20kV (Pulse)
1000X
Accuracy: ≤3%
Provides isolation between standard oscilloscope channels, isolation

Isolated front end

ISFE

betweenthe measured signal and ground. Uses USB 5V power supply, plug andplay.
The maximum input voltage allowed is up to ± 600Vpk.

STB-3 Test
Demo Board

USB Isolated AWG
ModuleRack Mount

WWW.SIGLENT.COM
15

Board

Output signals including square, sine, AM, fast edge, pulse, PWM, I2C,
CAN, LIN etc. Used in teaching and demonstrations.

Output Sine, Square, Ramp, pulse, Noise, DC and 45 built-in
SAG1021I

waveforms. The arbitrary waveforms can be accessed and edited by
the EasyWave PC software.

SDS1X-E-RMK

The height is 4U.

SDS2000X-E Series Digital Oscilloscope

Ordering Information
Ordering information
Product NameSDS2202X-E 200MHz Two Channels
SDS2352X-E 350MHz Two Channels

Standard Accessories

USB Cable -1
Quick Start -1
Passive Probe -2
Certification -1
Power Cord -1

Optional Accessories

16 Channels Logic Analyzer

SLA1016

USB Isolated AWG Module Hardware

SAG1021I

Isolated Front EndISFESTB Demo Source

STB-3

High Voltage Probe

HPB4010CP4020/CP4050/CP4070/

Current Probes

CP4070A/CP5030/CP5030A/
CP5150/CP5500

Differential Probes
Rack Mount

16
WWW.SIGLENT.COM

DPB4080/DPB5150/DPB5150A/DPB5700/DPB5700A
SDS1X-E-RMK

WWW.SIGLENT.COM
17