SDS6000L Series
Low Profile
Digital Oscilloscope
User Manual
EN01A

SDS6000L User ManualContents
CONTENTS ............................................................................................................................. 1
1

INTRODUCTION ................................................................................................................ 92

IMPORTANT SAFETY INFORMATION ............................................................................ 10
2.1

GENERAL SAFETY SUMMARY ................................................................................................... 10

2.2

SAFETY TERMS AND SYMBOLS ................................................................................................ 13

2.3

WORKING ENVIRONMENT ........................................................................................................ 14

2.4

COOLING REQUIREMENTS ....................................................................................................... 15

2.5

POWER AND GROUNDING REQUIREMENTS ............................................................................... 16

2.6

CLEANING ............................................................................................................................... 17

2.7

ABNORMAL CONDITIONS.......................................................................................................... 17

2.8

SAFETY COMPLIANCE .............................................................................................................. 18

INFORMATIONS ESSENTIELLES SUR LA SECURITE ....................................................... 19
EXIGENCE DE SECURITE ................................................................................................................... 19
TERMES ET SYMBOLES DE SECURITE ................................................................................................. 21
ENVIRONNEMENT DE TRAVAIL ............................................................................................................ 22
EXIGENCES DE REFROIDISSEMENT .................................................................................................... 24
CONNEXIONS D'ALIMENTATION ET DE TERRE ...................................................................................... 24
NETTOYAGE ..................................................................................................................................... 25
CONDITIONS ANORMALES.................................................................................................................. 26
CONFORMITE EN MATIERE DE SECURITE ............................................................................................ 26

3

FIRST STEPS .................................................................................................................. 27
3.1

DELIVERY CHECKLIST ............................................................................................................. 27

3.2

QUALITY ASSURANCE .............................................................................................................. 27

3.3

MAINTENANCE AGREEMENT .................................................................................................... 27

4

DOCUMENT CONVENTIONS.......................................................................................... 28

5

GETTING STARTED ........................................................................................................ 29
5.1

MECHANICAL DIMENSION ........................................................................................................ 29

5.2

FRONT PANEL OVERVIEW ........................................................................................................ 30

5.3

REAR PANEL OVERVIEW.......................................................................................................... 31

5.4

TO INSTALL THE RACKMOUNT FLANGE KIT ............................................................................... 32

5.5

CONNECTING TO EXTERNAL DEVICES/SYSTEMS....................................................................... 32

5.5.1

Power Supply ........................................................................................................................ 32

5.5.2

Probes ................................................................................................................................... 32

5.5.3

LAN ....................................................................................................................................... 34

5.5.4

External Monitor and Mouse ................................................................................................. 34

5.5.5

Auxiliary Output ..................................................................................................................... 34

int.siglent.com

1

SDS6000L User Manual

6

7

8

9

5.5.6

Reference Input and Output .................................................................................................. 35

5.5.7

Waveform Generator ............................................................................................................. 35

5.5.8

Logic Probe ........................................................................................................................... 35

5.6

POWER ON ............................................................................................................................. 36

5.7

SHUT DOWN ............................................................................................................................ 36

5.8

SYSTEM INFORMATION ............................................................................................................ 37

5.9

INSTALL OPTIONS .................................................................................................................... 37

REMOTE CONTROL ....................................................................................................... 38
6.1

WEB BROWSER ...................................................................................................................... 38

6.2

OTHER CONNECTIVITY ............................................................................................................ 39

SCREEN DISPLAY .......................................................................................................... 40
7.1

OVERVIEW .............................................................................................................................. 40

7.2

MENU BAR .............................................................................................................................. 41

7.3

GRID AREA ............................................................................................................................. 41

7.4

CHANNEL DESCRIPTOR BOX .................................................................................................... 43

7.5

TIMEBASE AND TRIGGER DESCRIPTOR BOXES ......................................................................... 44

7.6

DIALOG BOX ........................................................................................................................... 47

7.7

MOUSE CONTROL ................................................................................................................... 49

7.8

CHOOSING THE LANGUAGE...................................................................................................... 50

MULTIPLE APPROACHES TO RECALL FUNCTIONS ................................................... 51
8.1

MENU BAR .............................................................................................................................. 51

8.2

DESCRIPTOR BOX ................................................................................................................... 51

VERTICAL SETUP .......................................................................................................... 52
9.1

TURN ON/OFF A CHANNEL ........................................................................................................ 52

9.2

CHANNEL SETUP ..................................................................................................................... 53

10 DIGITAL CHANNELS ...................................................................................................... 60
10.1 OVERVIEW .............................................................................................................................. 60
10.2 ENABLE/DISABLE THE DIGITAL CHANNELS ................................................................................ 61
10.3 DIGITAL CHANNEL SETUP ........................................................................................................ 62

11 HORIZONTAL AND ACQUISITION SETUP ..................................................................... 66
11.1 TIMEBASE SETUP .................................................................................................................... 66
11.2 ACQUISITION SETUP................................................................................................................ 67
11.2.1 Overview ............................................................................................................................... 67
11.2.2 Acquisition ............................................................................................................................. 69
11.2.3 Memory Management ........................................................................................................... 72
11.2.4 Roll Mode .............................................................................................................................. 73
11.2.5 Sequence .............................................................................................................................. 73
11.2.6 ESR ....................................................................................................................................... 76

11.3 HISTORY ................................................................................................................................. 80
2

int.siglent.com

SDS6000L User Manual

12 ZOOM .............................................................................................................................. 83
13 TRIGGER ......................................................................................................................... 85
13.1 OVERVIEW .............................................................................................................................. 85
13.2 TRIGGER SETUP ..................................................................................................................... 86
13.3 TRIGGER LEVEL ...................................................................................................................... 87
13.4 TRIGGER MODE ...................................................................................................................... 88
13.5 TRIGGER TYPE........................................................................................................................ 89
13.5.1 Overview ............................................................................................................................... 89
13.5.2 Edge Trigger .......................................................................................................................... 90
13.5.3 Slope Trigger ......................................................................................................................... 90
13.5.4 Pulse Trigger ......................................................................................................................... 92
13.5.5 Video Trigger ......................................................................................................................... 94
13.5.6 Window Trigger ..................................................................................................................... 98
13.5.7 Interval Trigger ...................................................................................................................... 99
13.5.8 Dropout Trigger ................................................................................................................... 100
13.5.9 Runt Trigger ........................................................................................................................ 101
13.5.10 Pattern Trigger .................................................................................................................... 101
13.5.11 Qualified Trigger .................................................................................................................. 103
13.5.12 Nth Edge Trigger ................................................................................................................. 104
13.5.13 Delay Trigger ....................................................................................................................... 105
13.5.14 Setup/Hold Trigger .............................................................................................................. 105
13.5.15 Serial Trigger ....................................................................................................................... 106

13.6 TRIGGER SOURCE................................................................................................................. 106
13.7 HOLDOFF .............................................................................................................................. 107
13.8 TRIGGER COUPLING .............................................................................................................. 108
13.9 NOISE REJECT ...................................................................................................................... 109
13.10 ZONE TRIGGER ..................................................................................................................... 110

14 SERIAL TRIGGER AND DECODE ................................................................................ 115
14.1 OVERVIEW ............................................................................................................................ 115
14.2 I2C TRIGGER AND SERIAL DECODE ....................................................................................... 117
14.2.1 I2C Signal Settings.............................................................................................................. 117
14.2.2 I2C Trigger .......................................................................................................................... 118
14.2.3 I2C Serial Decode ............................................................................................................... 122

14.3 SPI TRIGGER AND SERIAL DECODE ....................................................................................... 125
14.3.1 SPI Signal Settings ............................................................................................................. 125
14.3.2 SPI Trigger .......................................................................................................................... 128
14.3.3 SPI Serial Decode ............................................................................................................... 128

14.4 UART TRIGGER AND SERIAL DECODE ................................................................................... 129
14.4.1 UART Signal Settings ......................................................................................................... 129
int.siglent.com

3

SDS6000L User Manual

14.4.2 UART Trigger ...................................................................................................................... 130
14.4.3 UART Serial Decode ........................................................................................................... 130

14.5 CAN TRIGGER AND SERIAL DECODE ..................................................................................... 131
14.5.1 CAN Signal Settings............................................................................................................ 131
14.5.2 CAN Trigger ........................................................................................................................ 131
14.5.3 CAN Serial Decode ............................................................................................................. 132

14.6 LIN TRIGGER AND SERIAL DECODE ....................................................................................... 134
14.6.1 LIN Signal Settings.............................................................................................................. 134
14.6.2 LIN Trigger .......................................................................................................................... 134
14.6.3 LIN Serial Decode ............................................................................................................... 135

14.7 FLEXRAY TRIGGER AND SERIAL DECODE ............................................................................... 136
14.7.1 FlexRay Signal Settings ...................................................................................................... 136
14.7.2 FlexRay Trigger ................................................................................................................... 136
14.7.3 FlexRay Serial Decode ....................................................................................................... 137

14.8 CAN FD TRIGGER AND SERIAL DECODE ................................................................................ 139
14.8.1 CAN FD Signal Settings ...................................................................................................... 139
14.8.2 CAN FD Trigger ................................................................................................................... 139
14.8.3 CAN FD Serial Decode ....................................................................................................... 140

14.9 I2S TRIGGER AND SERIAL DECODE ........................................................................................ 142
14.9.1 I2S Signal Settings .............................................................................................................. 142
14.9.2 I2S Trigger ........................................................................................................................... 143
14.9.3 I2S Serial Decode ............................................................................................................... 144

14.10 MIL-STD-1553B TRIGGER AND SERIAL DECODE ................................................................... 145
14.10.1 MIL-STD-1553B Signal Settings ......................................................................................... 145
14.10.2 MIL-STD-1553B Serial Decode .......................................................................................... 145

14.11 SENT TRIGGER AND SERIAL DECODE ................................................................................... 146
14.11.1 SENT Signal Settings.......................................................................................................... 146
14.11.2 SENT Trigger ...................................................................................................................... 147
14.11.3 SENT Serial Decode ........................................................................................................... 150

14.12 MANCHESTER SERIAL DECODE ............................................................................................. 151
14.12.1 Manchester Signal Settings ................................................................................................ 152
14.12.2 Manchester Serial Decode .................................................................................................. 153

15 CURSORS ..................................................................................................................... 154
15.1 OVERVIEW ............................................................................................................................ 154
15.2 SELECT AND MOVE CURSORS ............................................................................................... 160

16 MEASUREMENT ........................................................................................................... 162
16.1 OVERVIEW ............................................................................................................................ 162
16.2 SET PARAMETERS................................................................................................................. 163
16.3 TYPE OF MEASUREMENT ....................................................................................................... 166
4

int.siglent.com

SDS6000L User Manual

16.3.1 Vertical Measurement ......................................................................................................... 166
16.3.2 Horizontal Measurement ..................................................................................................... 168
16.3.3 Miscellaneous Measurements ............................................................................................ 169
16.3.4 Delay Measurement ............................................................................................................ 170

16.4 TREND .................................................................................................................................. 171
16.5 TRACK .................................................................................................................................. 172
16.6 DISPLAY MODE ..................................................................................................................... 173
16.7 MEASUREMENT STATISTICS ................................................................................................... 174
16.8 STATISTICS HISTOGRAM ........................................................................................................ 175
16.9 SIMPLE MEASUREMENTS ....................................................................................................... 176
16.10 GATE .................................................................................................................................... 177
16.11 AMPLITUDE STRATEGY .......................................................................................................... 178
16.12 THRESHOLD .......................................................................................................................... 178
16.13 HARDWARE FREQUENCY COUNTER ....................................................................................... 179

17 MATH ............................................................................................................................. 180
17.1 OVERVIEW ............................................................................................................................ 180
17.2 ARITHMETIC .......................................................................................................................... 181
17.2.1 Addition / Subtraction / Multiplication / Division .................................................................. 181
17.2.2 Identity / Negation ............................................................................................................... 182
17.2.3 Average / ERES .................................................................................................................. 183
17.2.4 Max-hold / Min-hold............................................................................................................. 183

17.3 ALGEBRA .............................................................................................................................. 183
17.3.1 Differential ........................................................................................................................... 183
17.3.2 Integral ................................................................................................................................ 184
17.3.3 Square Root ........................................................................................................................ 185
17.3.4 Absolute .............................................................................................................................. 186
17.3.5 Sign ..................................................................................................................................... 186
17.3.6 exp/exp10 ............................................................................................................................ 187
17.3.7 ln/lg ...................................................................................................................................... 187
17.3.8 Interpolate ........................................................................................................................... 188

17.4 FILTER .................................................................................................................................. 188
17.5 FREQUENCY ANALYSIS .......................................................................................................... 190
17.6 FORMULA EDITOR ................................................................................................................. 198

18 REFERENCE ................................................................................................................. 200
19 MEMORY ....................................................................................................................... 202
20 SEARCH ........................................................................................................................ 204
21 NAVIGATE ..................................................................................................................... 207
22 MASK TEST .................................................................................................................. 212
22.1 OVERVIEW ............................................................................................................................ 212
int.siglent.com

5

SDS6000L User Manual

22.2 MASK SETUP ........................................................................................................................ 214
22.2.1 Create Mask ........................................................................................................................ 214
22.2.2 Mask Editor ......................................................................................................................... 215

22.3 PASS/FAIL RULE ................................................................................................................... 217
22.4 OPERATION ........................................................................................................................... 217

23 DVM ............................................................................................................................... 218
23.1 OVERVIEW ............................................................................................................................ 218
23.2 MODE ................................................................................................................................... 219
23.3 DIAGRAMS ............................................................................................................................ 220

24 COUNTER ..................................................................................................................... 223
24.1 OVERVIEW ............................................................................................................................ 223
24.2 MODE ................................................................................................................................... 225

25 HISTOGRAM ................................................................................................................. 226
25.1 OVERVIEW ............................................................................................................................ 226
25.2 REGION SETTING .................................................................................................................. 228

26 POWER ANALYSIS ....................................................................................................... 230
26.1 OVERVIEW ............................................................................................................................ 230
26.2 POWER QUALITY ................................................................................................................... 230
26.3 CURRENT HARMONICS .......................................................................................................... 233
26.4 INRUSH CURRENT ................................................................................................................. 235
26.5 SWITCHING LOSS .................................................................................................................. 236
26.6 SLEW RATE........................................................................................................................... 239
26.7 MODULATION ........................................................................................................................ 240
26.8 OUTPUT RIPPLE .................................................................................................................... 240
26.9 TURN ON/TURN OFF ............................................................................................................. 241
26.10 TRANSIENT RESPONSE.......................................................................................................... 242
26.11 PSRR .................................................................................................................................. 244
26.12 POWER EFFICIENCY .............................................................................................................. 245
26.13 SOA..................................................................................................................................... 245

27 BODE PLOT .................................................................................................................. 248
27.1 OVERVIEW ............................................................................................................................ 248
27.2 CONFIGURATION ................................................................................................................... 249
27.2.1 Connection .......................................................................................................................... 249
27.2.2 Sweep ................................................................................................................................. 249

27.3 DISPLAY................................................................................................................................ 252
27.4 DATA ANALYSIS ..................................................................................................................... 254

28 EYE DIAGRAM .............................................................................................................. 257
28.1 OVERVIEW ............................................................................................................................ 257

6

int.siglent.com

SDS6000L User Manual

28.2 SIGNAL SETTING ................................................................................................................... 259
28.3 CLOCK RECOVERY ................................................................................................................ 260
28.4 MEASUREMENT ..................................................................................................................... 261
28.5 MASK TEST........................................................................................................................... 263
28.6 OTHER OPERATION ............................................................................................................... 263

29 JITTER ANALYSIS ........................................................................................................ 264
29.1 OVERVIEW ............................................................................................................................ 264
29.2 SIGNAL CONFIGURATION ....................................................................................................... 265
29.3 CLOCK RECOVERY ................................................................................................................ 266
29.4 JITTER DECOMPOSITION........................................................................................................ 266
29.5 JITTER MEASURE .................................................................................................................. 267
29.6 OTHER OPERATION ............................................................................................................... 270
29.7 SYSTEM EFFECT ON JITTER MEASURE ................................................................................... 270

30 DISPLAY ........................................................................................................................ 271
31 WAVEFORM GENERATOR ........................................................................................... 279
31.1 OVERVIEW ............................................................................................................................ 279
31.2 WAVE TYPE .......................................................................................................................... 280
31.3 OTHER SETTING ................................................................................................................... 281
31.4 SYSTEM ................................................................................................................................ 282

32 SAVE/RECALL .............................................................................................................. 284
32.1 SAVE TYPE ........................................................................................................................... 284
32.2 FILE MANAGER ..................................................................................................................... 287
32.3 SAVE AND RECALL INSTANCES ............................................................................................... 288

33 UTILITY ......................................................................................................................... 292
33.1 SYSTEM INFORMATION .......................................................................................................... 292
33.2 SYSTEM SETTING .................................................................................................................. 292
33.2.1 Language ............................................................................................................................ 292
33.2.2 Screen Saver ...................................................................................................................... 293
33.2.3 Sound .................................................................................................................................. 293
33.2.4 Auto Power-on .................................................................................................................... 293
33.2.5 Date/Time ............................................................................................................................ 293
33.2.6 Reference Position Setting ................................................................................................. 294
33.2.7 Tips...................................................................................................................................... 297

33.3 I/O SETTING ......................................................................................................................... 297
33.3.1 LAN ..................................................................................................................................... 297
33.3.2 Clock Source ....................................................................................................................... 298

33.4 INSTALL OPTIONS .................................................................................................................. 299
33.5 MAINTENANCE ...................................................................................................................... 300
33.5.1 Upgrade .............................................................................................................................. 300
int.siglent.com

7

SDS6000L User Manual

33.5.2 Self-Calibration.................................................................................................................... 301
33.5.3 Developer Options .............................................................................................................. 302

33.6 SERVICE ............................................................................................................................... 302
33.6.1 Web ..................................................................................................................................... 302
33.6.2 Network Mapping ................................................................................................................ 302
33.6.3 Emulation ............................................................................................................................ 303
33.6.4 LXI ....................................................................................................................................... 304
33.6.5 Share File ............................................................................................................................ 305

34 TROUBLESHOOTING ................................................................................................... 306

8

int.siglent.com

SDS6000L User Manual

1 Introduction
A digital oscilloscope is a multi-functional instrument for displaying, analyzing, and storing electricalsignals. It is an indispensable tool for designing, manufacturing, and maintaining electronic equipment.
This user manual includes important safety and installation information related to the SDS6000L seriesof low-profile oscilloscopes and includes simple tutorials for the basic operation of the instrument.

The series includes the following models:Model

AnalogBandwidth

Maximum Sampling Rate

AnalogChannels

SDS6208L

2 GHz

5 GSa/s (10 GSa/s ESR)
@ each channel

8

SDS6204L

2 GHz

5 GSa/s (10 GSa/s ESR)
@ each channel

4

SDS6108L

1 GHz

5 GSa/s (10 GSa/s ESR)
@ each channel

8

SDS6104L

1 GHz

5 GSa/s (10 GSa/s ESR)
@ each channel

4

SDS6058L

500 MHz

5 GSa/s (10 GSa/s ESR)
@ each channel

8

SDS6054L

500 MHz

5 GSa/s (10 GSa/s ESR)
@ each channel

4

int.siglent.com

9

SDS6000L User Manual2

Important Safety Information

This manual contains information and warnings that must be followed by the user for safe operationand to keep the product in a safe condition.

2.1

General Safety Summary

Carefully read the following safety precautions to avoid personal injury and prevent damage to the
instrument and any products connected to it. To avoid potential hazards, please use the instrument as
specified.

To Avoid Fire or Personal Injury.

Use the Proper Power Line.
Only use a local / state-approved power cord for connecting the instrument to mains power sources.

Ground the Instrument.
The instrument grounds through the protective terra conductor of the power line. To avoid electric
shock, the ground conductor must be connected to the earth. Make sure the instrument is grounded
correctly before connecting its input or output terminals.

Connect the Signal Wire Correctly.
The potential of the signal wire is equal to the earth, so do not connect the signal wire to a high voltage.
Do not touch the exposed contacts or components.

Look over All Terminals’ Ratings.
To avoid fire or electric shock, please look over all ratings and signed instructions of the instrument.
Before connecting the instrument, please read the manual carefully to gain more information about the
ratings.

Equipment Maintenance and Service.
When the equipment fails, please do not dismantle the machine for maintenance. The equipment
10

int.siglent.com

SDS6000L User Manualcontains capacitors, power supply, transformers, and other energy storage devices, which may cause
high voltage damage. The internal devices of the equipment are sensitive to static electricity, and directcontact can easily cause irreparable damage to the equipment. It is necessary to return to the factoryor the company's designated maintenance organization for maintenance. Be sure to pull out the power
supply when repairing the equipment. Live line operation is strictly prohibited. The equipment can only
be powered on when the maintenance is completed and the maintenance is confirmed to be successful.

Identification of Normal State of Equipment.
After the equipment is started, there will be no alarm information or error information at the interface
under normal conditions. If there is a pop-up window or button during the scanning process or there isan alarm or error prompt, the device may be in an abnormal state. You need to view the specific prompt
information. You can try to restart the instrument to see if it corrects the error condition. If the fault
information is still in place, do not use the instrument for testing. Contact the manufacturer or the
maintenance department designated by the manufacturer to carry out maintenance before reusing the
product.

Not Operate with Suspected Failures.
If you suspect that there is damage to the instrument, please let qualified service personnel check it.

Avoid Circuit or Wire Exposed Components Exposed.
Do not touch exposed contacts or components when the power is on.

Do not operate in wet/damp conditions.

Do not operate in an explosive atmosphere.

Keep the surface of the instrument clean and dry.

Only probe assemblies that meet the requirement of UL61010-031 and CAN/CSA-C22.2
No.61010-031 shall be used.

Only a lithium battery with the same specifications as the original battery should be used toint.siglent.com

11

SDS6000L User Manualreplace the battery on the mainboard.

Do not use the equipment for measurements on mains circuits. Do not use the equipment formeasurements on voltages exceeding the voltage ranges described in the manual. The
maximum additional transient voltage cannot exceed 1300 V.

The responsible body or operator should refer to the instruction manual to preserve theprotection afforded by the equipment. If the equipment is used in a manner not specified by the
manufacturer, the protection provided by the equipment may be impaired.

Any parts of the device and its accessories are not allowed to be changed or replaced, otherthan authorized by the manufacturer or agent.

12

int.siglent.com

SDS6000L User Manual

2.2

Safety Terms and Symbols

When the following symbols or terms appear on the front or rear panel of the instrument or in thismanual, they indicate special care in terms of safety.

This symbol is used where caution is required. Refer to the accompanying
information or documents to protect against personal injury or damage to the
instrument.

This symbol warns of a potential risk of shock hazard.

This symbol is used to denote the measurement ground connection.

This symbol is used to denote a safety ground connection.

This symbol shows that the switch is an On/Standby switch. When it is pressed,
the scope’s state switches between Operation and Standby. This switch does notdisconnect the device's power supply. To completely power off the scope, thepower switch on the rear panel should be turned to “Off”.

This symbol is used to represent alternating current, or "AC".

The "CAUTION" symbol indicates a potential hazard. It calls attention to a
CAUTION

procedure, practice, or condition which may be dangerous if not followed. Do notproceed until its conditions are fully understood and met.
The "WARNING" symbol indicates a potential hazard. It calls attention to a

WARNINGprocedure, practice, or condition which, if not followed, could cause bodily injury
or death. If a WARNING is indicated, do not proceed until the safety conditions
are fully understood and met.

int.siglent.com

13

SDS6000L User Manual

2.3

Working Environment

The design of the instrument has been verified to conform to EN 61010-1 safety standard per thefollowing limits:
Environment
The instrument is used indoors and should be operated in a clean and dry environment with an ambient
temperature range.

Note: Direct sunlight, electric heaters, and other heat sources should beconsidered when evaluating the ambient temperature.

Warning: Do not operate the instrument in explosive, dusty, or humid
environments.

Ambient Temperature
Operating: 0 ℃ to +50 ℃
Non-operation: -30 ℃ to +70 ℃

Note: Direct sunlight, radiators, and other heat sources should be taken into
account when assessing the ambient temperature.

Humidity
Operating: 5% ~ 90% RH, 30 ℃, derate to 50% RH at 40 ℃
Non-operating: 5% ~ 95% RH

AltitudeOperating: ≤ 3,048 m, 25 ℃
Non-operating: ≤ 12,191 m

14

int.siglent.com

SDS6000L User Manual

Installation (overvoltage) Category
This product is powered by mains conforming to installation (overvoltage) Category II.

Note: Installation (overvoltage) category I refers to situations where
equipment measurement terminals are connected to the source circuit. In
these terminals, precautions are done to limit the transient voltage to acorrespondingly low level.
Installation (overvoltage) category II refers to the local power distribution level
which applies to equipment connected to the AC line (AC power).

Degree of Pollution
The oscilloscopes may be operated in environments of Pollution Degree II.

Note: Degree of Pollution II refers to a working environment that is dry andnon-conductive pollution occurs. Occasional temporary conductivity causedby condensation is expected.

IP Rating
IP20 (as defined in IEC 60529).

2.4

Cooling Requirements

This instrument relies on forced air cooling with internal fans and ventilation openings. Care must be
taken to avoid restricting the airflow around the apertures (fan holes) on each side of the scope. To
ensure adequate ventilation it is required to leave a 15 cm (6 inch) minimum gap around the sides of
the instrument.

CAUTION: Do not block the ventilation holes located on both sides of thescope.

int.siglent.com

15

SDS6000L User Manual

CAUTION: Do not allow any foreign matter to enter the scope through theventilation holes, etc.

2.5Power and Grounding Requirements

The instrument operates with a single-phase, 100 to 240 Vrms (+/-10%) AC power at 50/60 Hz (+/5%), or single-phase 100 to 120 Vrms (+/-10%) AC power at 400 Hz (+/-5%).

No manual voltage selection is required because the instrument automatically adapts to line voltage.

Depending on the type and number of options and accessories (probes, PC port plug-in, etc.), theinstrument can consume up to 380 W of power for the 8-channel models and 193 W for the 4-channel
models.

The instrument automatically adapts to the AC line input within the following ranges:
Voltage Range:

90 - 264 Vrms

90 - 132 Vrms

Frequency Range:

47 - 63 Hz

380 - 420 Hz

The instrument includes a grounded cord set containing a molded three-terminal polarized plug and astandard IEC320 (Type C13) connector for making line voltage and safety ground connections. The
AC inlet ground terminal is connected directly to the frame of the instrument. For adequate protection
against electrical shock hazards, the power cord plug must be inserted into a mating AC outlet
containing a safety ground contact. Use only the power cord specified for this instrument and certified
for the country of use.

Warning: Electrical Shock Hazard!
Any interruption of the protective conductor inside or outside of thescope, or disconnection of the safety ground terminal creates ahazardous situation.
Intentional interruption is prohibited.

16

int.siglent.com

SDS6000L User Manual

The position of the oscilloscope should allow easy access to the socket. To make the oscilloscopecompletely power off, unplug the instrument power cord from the AC socket.

The power cord should be unplugged from the AC outlet if the scope is not to be used for an extended
period.

CAUTION: The outer shells of the front panel terminals (C1~C8, EXT) areconnected to the instrument’s chassis and therefore to the safety ground.

2.6

Cleaning

Clean only the exterior of the instrument, using a damp, soft cloth. Do not use chemicals or abrasive
elements. Under no circumstances allow moisture to penetrate the instrument. To avoid electricalshock, unplug the power cord from the AC outlet before cleaning.

Warning: Electrical Shock Hazard!
No operator serviceable parts inside. Do not remove covers.
Refer servicing to qualified personnel

2.7

Abnormal Conditions

Do not operate the scope if there is any visible sign of damage or has been subjected to severetransport stresses.

If you suspect the scope’s protection has been impaired, disconnect the power cord and secure theinstrument against any unintended operation.

Proper use of the instrument depends on the careful reading of all instructions and labels.

int.siglent.com

17

SDS6000L User Manual

Warning: Any use of the scope in a manner not specified by the manufacturer
may impair the instrument’s safety protection. This instrument should not bedirectly connected to human subjects or used for patient monitoring.

2.8

Safety ComplianceThis section lists the safety standards with which the product complies.

U.S. nationally recognized testing laboratory listing

UL 61010-1:2012/R: 2018-11. Safety Requirements for Electrical Equipment for Measurement,
Control, and Laboratory Use – Part 1: General Requirements.



UL 61010-2-030:2018. Safety Requirements for Electrical Equipment for Measurement, Control,and Laboratory Use – Part2-030: Particular requirements for testing and measuring circuits.

Canadian certification


CAN/CSA-C22.2 No. 61010-1:2012/A1:2018-11. Safety Requirements for Electrical Equipment
for Measurement, Control, and Laboratory Use – Part 1: General Requirements.



CAN/CSA-C22.2 No. 61010-2-030:2018. Safety Requirements for Electrical Equipment for
Measurement, Control, and Laboratory Use – Part 2-030: Particular requirements for testing andmeasuring circuits.

18

int.siglent.com

SDS6000L User Manual

Informations essentielles sur la sécurité
Ce manuel contient des informations et des avertissements que les utilisateurs doivent suivre pour
assurer la sécurité des opérations et maintenir les produits en sécurité.

Exigence de Sécurité
Lisez attentivement les précautions de sécurité ci - après afin d 'éviter les dommages corporels et deprévenir les dommages aux instruments et aux produits associés. Pour éviter les risques potentiels,
utilisez les instruments prescrits.

Éviter l 'incendie ou les lésions corporelles.

Utilisez un cordon d'alimentation approprié.
N'utilisez que des cordons d'alimentation spécifiques aux instruments approuvés par les autorités
locales.

Mettez l'instrument au sol.
L'instrument est mis à la Terre par un conducteur de mise à la terre de protection du cordon
d'alimentation.Pour éviter un choc électrique, le conducteur de mise à la terre doit être mis à laterre.Assurez - vous que l'instrument est correctement mis à la terre avant de connecter les bornes
d'entrée ou de sortie de l'instrument.

Connectez correctement le fil de signalisation.
Le potentiel de la ligne de signal est égal au potentiel au sol, donc ne connectez pas la ligne de signal
à haute tension.Ne touchez pas les contacts ou les composants exposés.

Voir les cotes de tous les terminaux.
Pour éviter un incendie ou un choc électrique, vérifiez toutes les cotes et signez les instructions del'instrument.Avant de brancher l'instrument, lisez attentivement ce manuel pour obtenir de plus amples
renseignements sur les cotes.

int.siglent.com

19

SDS6000L User Manual

Entretien du matériel.
En cas de défaillance de l'équipement, ne pas démonter et entretenir l'équipement sans autorisation.
L'équipement contient des condensateurs, de l'alimentation électrique, des transformateurs et d'autres
dispositifs de stockage d'énergie, ce qui peut causer des blessures à haute tension. Les dispositifs
internes de l'équipement sont sensibles à l'électricité statique. Le contact direct peut facilement causer
des blessures irrécupérables à l'équipement. L'équipement doit être retourné àl'usine ou à l'organismede maintenance désigné par l'entreprise pour l'entretien. L'alimentation électrique doit être retirée
pendant l'entretienLa ligne ne doit pas être mise sous tension tant que l'entretien de l'équipement n'est
pas terminé et que l'entretien n'est pas confirmé.

Identification de l'état normal de l'équipement.
Après le démarrage de l'équipement, dans des conditions normales, il n'y aura pas d'informationd'alarme et d'erreur au bas de l'interface, et la courbe de l'interface sera balayée librement de gaucheà droite; si un blocage se produit pendant le processus de numérisation, ou si l'information d'alarmeou d'erreur apparaît au bas de l'interface, l'équipement peut être dans un état anormal. Pour voir
l'information d'alarme spécifique, vous pouvez d'abord essayer de redémarrerSi l'information sur ladéfaillance est toujours présente, ne l'utilisez pas pour l'essai. Contactez le fabricant ou le Service deréparation désignépar le fabricant pour effectuer l'entretien afin d'éviter d'apporter des données d'essai
erronées ou de mettre en danger la sécurité personnelle en raison de l'utilisation de la défaillance.

Ne pas fonctionner en cas de suspicion de défaillance.
Si vous soupçonnez des dommages à l'instrument, demandez à un technicien qualifié de vérifier.

L 'exposition du circuit ou de l' élément d 'exposition du fil est évitée.
Lorsque l 'alimentation est connectée, aucun contact ou élément nu n' est mis en contact.

Ne pas fonctionner dans des conditions humides / humides.

Pas dans un environnement explosif.

Maintenez la surface de l 'instrument propre et sec.

20

int.siglent.com

SDS6000L User Manual

Le Circuit d 'alimentation électrique ne peut pas être mesuré à l' aide du dispositif, ni la tension
qui dépasse la plage de tension décrite dans le présent manuel.

Seuls les ensembles de sondes conformes aux spécifications du fabricant peuvent être utilisés.

L'organisme ou l'opérateur responsable doit se référer au cahier des charges pour protéger laprotection offerte par le matériel.La protection offerte par le matériel peut être compromise si
celui - ci est utilisé de manière non spécifiée par le fabricant.

Aucune pièce du matériel et de ses annexes ne peut être remplacée ou remplacée sansl'autorisation de son fabricant.

Remplacer la batterie dans l 'appareil avec les mêmes spécifications de batterie au lithium.

Termes et symboles de sécurité
Lorsque les symboles ou termes suivants apparaissent sur le panneau avant ou arrière de l'instrument
ou dans ce manuel, ils indiquent un soin particulier en termes de sécurité.

Ce symbole est utilisé lorsque la prudence est requise. Reportez-vous aux
informations ou documents joints afin de vous protéger contre les blessures ou lesdommages à l'instrument.
Ce symbole avertit d'un risque potentiel de choc électrique.

Ce symbole est utilisé pour désigner la connexion de terre de mesure.

Ce symbole est utilisé pour indiquer une connexion à la terre de sécurité.
Ce symbole indique que l'interrupteur est un interrupteur marche / veille. Lorsqu'il
est enfoncé, l'état de l'oscilloscope bascule entre Fonctionnement et Veille. Cecommutateur ne déconnecte pas l'alimentation de l'appareil. Pour éteindrecomplètement l'oscilloscope, le cordon d'alimentation doit être débranché de laprise secteur une fois l'oscilloscope en état de veille.

int.siglent.com

21

SDS6000L User Manual

Ce symbole est utilisé pour représenter un courant alternatif, ou "AC".

CAUTION

Le symbole " CAUTION" indique un danger potentiel. Il attire l'attention sur une
procédure, une pratique ou une condition qui peut être dangereuse si elle n'est pas
suivie. Ne continuez pas tant que ses conditions n'ont pas été entièrement
comprises et remplies.

WARNINGLe symbole " WARNING" indique un danger potentiel. Il attire l'attention sur une
procédure, une pratique ou une condition qui, si elle n'est pas suivie, pourrait
entraîner des blessures corporelles ou la mort. Si un AVERTISSEMENT est
indiqué, ne continuez pas tant que les conditions de sécurité ne sont pas
entièrement comprises et remplies.

Environnement de travail
La conception de l'instrument a été certifiée conforme à la norme EN 61010-1, sur la base des valeurs
limites suivantes:
Environnement
L'instrument doit être utilisé à l'intérieur dans un environnement propre et sec dans la plage detempérature ambiante.

Note: la lumière directe du soleil, les réchauffeurs électriques et d'autres
sources de chaleur doivent être pris en considération lors de l'évaluation de latempérature ambiante.

Attention: ne pas utiliser l'instrument dans l'air explosif, poussiéreux ou
humide.

Température ambiante
En fonctionnement: 0 ℃ à +50 ℃
Hors fonctionnement: -30 ℃ à +70 ℃

Note: pour évaluer la température de l'environnement, il convient de tenir
compte des rayonnements solaires directs, des radiateurs thermiques etd'autres sources de chaleur.

22

int.siglent.com

SDS6000L User Manual

Humidité
Fonctionnement: 5% ~ 90% HR, 30 °C, 40 °C réduit à 50% HRHors fonctionnement: 5% ~ 95%, 65 ℃,
24 heures

Altitude
Fonctionnement: ≤ 3000 m
À l'arrêt: ≤ 12,191 m

Catégorie d 'installation (surtension)
Ce produit est alimenté par une alimentation électrique conforme à l 'installation (surtension)
Catégorie II.

Installation (overvoltage) Category Definitions Définition de catégorie d 'installation(surtension)
La catégorie II d'installation (surtension) est un niveau de signal applicable aux terminaux de mesured' équipement reliés au circuit source.Dans ces bornes, des mesures préventives sont prises pourlimiter la tension transitoire à un niveau inférieur correspondant.

La catégorie II d'installation (surtension) désigne le niveau local de distribution d 'énergie d' unéquipement conçu pour accéder à un circuit alternatif (alimentation alternative).

Degré de pollution
Un oscilloscope peut être utilisé dans un environnement Pollution Degree II.

Note: Pollution Degree II signifie que le milieu de travail est sec et qu'il y a une
pollution non conductrice.Parfois, la condensation produit une conductivité
temporaire.

IP Rating
IP20 (as defined in IEC 60529).

int.siglent.com

23

SDS6000L User Manual

Exigences de refroidissement
Cet instrument repose sur un refroidissement à air forcé avec des ventilateurs internes et desouvertures de ventilation. Des précautions doivent être prises pour éviter de restreindre le flux d'air
autour des ouvertures (trous de ventilateur) de chaque côté de la lunette. Pour assurer une ventilation
adéquate, il est nécessaire de laisser un espace minimum de 15 cm (6 pouces) sur les côtés del'instrument.

ATTENTION: Ne bloquez pas les trous de ventilation situés des deux côtésde la lunette.

ATTENTION: Ne laissez aucun corps étranger pénétrer dans la lunette parles trous de ventilation, etc.

Connexions d'alimentation et de terre
L'instrument fonctionne avec une alimentation CA monophasée de 100 à 240 Vrms (+/- 10%) à 50/60
Hz (+/- 5%), ou monophasée 100 - 120 Vrms (+/-10 %) Alimentation CA à 400 Hz (+/-5%).

Aucune sélection manuelle de la tension n'est requise car l'instrument s'adapte automatiquement à latension de ligne.
Selon le type et le nombre d'options et d'accessoires (sondes, plug-in de port PC, etc.), l'instrument
peut consommer jusqu'à 380 W de puissance pour les modèles à 8 canaux et 193 W pour les modèlesà 4 canaux.

Remarque: l'instrument s'adapte automatiquement à l'entrée de ligne CA dans les plages suivantes:
Plage de tension:

90 - 264 Vrms

90 - 132 Vrms

Gamme de fréquences:

47 - 63 Hz

380 - 420 Hz

L'instrument comprend un jeu de cordons mis à la terre contenant une fiche polarisée à trois bornes
moulée et un connecteur standard IEC320 (Type C13) pour établir la tension de ligne et la connexion
de mise à la terre de sécurité. La borne de mise à la terre de l'entrée CA est directement connectée24

int.siglent.com

SDS6000L User Manual

au châssis de l'instrument. Pour une protection adéquate contre les risques d'électrocution, la fiche ducordon d'alimentation doit être insérée dans une prise secteur correspondante contenant un contactde sécurité avec la terre. Utilisez uniquement le cordon d'alimentation spécifié pour cet instrument etcertifié pour le pays d'utilisation.

Avertissement: risque de choc électrique!
Toute interruption du conducteur de terre de protection à
l'intérieur ou à l'extérieur de la portée ou la déconnexion dela borne de terre de sécurité crée une situation dangereuse.
L'interruption intentionnelle est interdite.

La position de l'oscilloscope doit permettre un accès facile à la prise. Pour éteindre complètement
l'oscilloscope, débranchez le cordon d'alimentation de l'instrument de la prise secteur.

Le cordon d'alimentation doit être débranché de la prise secteur si la lunette ne doit pas être utilisée
pendant une période prolongée.

ATTENTION: les enveloppes extérieures des bornes du panneau avant(C1~C8, EXT) sont connectées au châssis de l'instrument et donc à la terrede sécurité.

Nettoyage
Nettoyez uniquement l'extérieur de l'instrument à l'aide d'un chiffon doux et humide. N'utilisez pas deproduits chimiques ou d'éléments abrasifs. Ne laissez en aucun cas l'humidité pénétrer dansl'instrument. Pour éviter les chocs électriques, débranchez le cordon d'alimentation de la prise secteur
avant de le nettoyer.

Avertissement: risque de choc électrique!
Aucune pièce réparable par l'opérateur à l'intérieur. Ne
retirez pas les capots.
Confiez l'entretien à un personnel qualifié

int.siglent.com

25

SDS6000L User Manual

Conditions anormales
Utilisez l'instrument uniquement aux fins spécifiées par le fabricant.
N'utilisez pas la lunette s'il y a des signes visibles de dommages ou si elle a été soumise à de fortes
contraintes de transport.

Si vous pensez que la protection de l'oscilloscope a été altérée, débranchez le cordon d'alimentationet sécurisez l'instrument contre toute opération involontaire.
Une bonne utilisation de l'instrument nécessite la lecture et la compréhension de toutes les instructionset étiquettes.

Avertissement: Toute utilisation de l'oscilloscope d'une manière non
spécifiée par le fabricant peut compromettre la protection de sécurité del'instrument. Cet instrument ne doit pas être directement connecté à dessujets humains ni utilisé pour la surveillance des patients.

Conformité en matière de sécurité
La présente section présente les normes de sécurité applicables aux produits.

U.S. nationally recognized testing laboratory listing■ UL 61010-1:2012/R:2018-11. Prescriptions en matière de sécurité pour les appareils électriques
utilisés en laboratoire et de mesure - partie 1: prescriptions générales.
■ UL 61010-2-030:2018. Prescriptions de sécurité pour les appareils électriques de mesure, decontrôle et de laboratoire - partie 2 - 030: prescriptions spéciales pour les circuits d 'essai et de mesure.

Canadian certification■ CAN/CSA-C22.2 No. 61010-1:2012/A1:2018-11. Prescriptions en matière de sécurité pour lesappareils électriques utilisés en laboratoire et de mesure - partie 1: prescriptions générales.
■ CAN/CSA-C22.2 No. 61010-2-030:2018. Prescriptions de sécurité pour les appareils électriques demesure, de contrôle et de laboratoire - partie 2 - 030: prescriptions spéciales pour les circuits d 'essai
et de mesure.

26

int.siglent.com

SDS6000L User Manual3
3.1First Steps
Delivery Checklist

First, verify that all items listed on the packing list have been delivered. If you note any omissions or
damage, please contact your nearest SIGLENT customer service center or distributor as soon as
possible. If you fail to contact us immediately in case of omission or damage, we will not be responsible
for replacement.

3.2

Quality Assurance

The oscilloscope has a 3-year warranty (1-year warranty for probe and accessories) from the date ofshipment, during normal use and operation. SIGLENT can repair or replace any product that is returnedto the authorized service center during the warranty period. We must first examine the product to makesure that the defect is caused by the process or material, not by abuse, negligence, accident, abnormal
conditions, or operation.

SIGLENT shall not be responsible for any defect, damage, or failure caused by any of the following:a) Attempted repairs or installations by personnel other than SIGLENT.
b) Connection to incompatible devices/incorrect connection.
c) For any damage or malfunction caused by the use of non-SIGLENT supplies. Furthermore,
SIGLENT shall not be obligated to service a product that has been modified. Spare,replacement parts and repairs have a 90-day warranty.
The oscilloscope's firmware has been thoroughly tested and is presumed to be functional. Nevertheless,
it is supplied without a warranty of any kind covering detailed performance. Products not made by
SIGLENT are covered solely by the warranty of the original equipment manufacturer.

3.3

Maintenance Agreement

We provide various services based on maintenance agreements. We offer extended warranties as wellas installation, training, enhancement and on-site maintenance, and other services through specialized
supplementary support agreements. For details, please consult your local SIGLENT customer service
center or distributor.

int.siglent.com

27

SDS6000L User Manual

4

Document Conventions

For convenience, italicized text with shading is used to represent the clickable menu/button/region on
the display. For example, Display represents the "Display" menu on the screen:

For the operations that contain multiple steps, the description is in the form of "Step 1 > Step 2 >...".
As an example, follow each step in the sequence to enter the upgrade interface:

Utility > Menu > Maintenance > Upgrade

Click the Utility menu on the menu bar as step 1, click the Menu option on the menu as step 2,
click the Maintenance option on the screen as step 3, and click the Update option on the screen
as step 4 to enter the upgrade interface.

28

int.siglent.com

SDS6000L User Manual

5
5.1

Getting Started
Mechanical Dimension

4-channel model

8-channel model

Front View

Top View

int.siglent.com

29

SDS6000L User Manual

5.2

Front Panel Overview

Analog Input Connectors 1 MΩ: ≤ 400 Vpk (DC + AC), DC~10 kHz; 50 Ω: ≤ 5 Vrms,±10 V PeakDigital Input Connector
USB 3.0 Host Ports Connect to USB storage devices for data transfer or USB
mouse/keyboard for control
Probe Compensation / Ground Terminal
Power Standby Button
Acquisition status and LAN status LEDs
Reset for LAN
Handles

30

int.siglent.com

SDS6000L User Manual

5.3

Rear Panel Overview

AC Power Input and Power Switch
SD Card Slot
HDMI Video Output Connect the port to an external monitor. The resolution is 1280 * 800
1000M LAN Port Connect the port to the network for remote control
USB 2.0 Hosts Connect with a USB storage device or USB mouse / keyboard
USB 2.0 Device Connects with a PC for remote control
Auxiliary Out Outputs the trigger indicator. When Mask Test is enabled, outputs the pass /fail signal
Ext Trigger Input
10 MHz Out and 10 MHz In
Built-in AWG Output

int.siglent.com

31

SDS6000L User Manual

To Install the Rackmount Flange Kit

5.4
1.

Install the handles to the left and right rack flange using four M4×8 screws.

2.

Install the rack flanges to the instrument using six M3×6 screws.

5.5
5.5.1

Connecting to External Devices/Systems
Power Supply

The standard power supply for the instrument is 100~240 V, 50/60 Hz, or 100~120 V, 400 Hz. Please
use the power cord provided with the instrument to connect it to AC power.

5.5.2

ProbesThe SDS6000L series oscilloscope supports active probes and passive probes. The specifications andprobe documents can be obtained at int.siglent.com, www.siglentna.com, or www.siglenteu.com.

32

int.siglent.com

SDS6000L User Manual

Probe Compensation
When a passive probe is used for the first time, you should compensate it to match the input channelof the oscilloscope. Non-compensated or poorly compensated probes may increase measurement
inaccuracy or error. The probe compensation procedures are as follows:

1. Connect the coaxial cable interface (BNC connector) of the passive probe to any channel ofthe oscilloscope.
2. Connect the probe to the “Compensation Signal Output Terminal” (Cal) on the front of theoscilloscope. Connect the ground alligator clip of the probe to the “Ground Terminal” under
the compensation signal output terminal.

3. Perform Acquire > Auto Setup .
4. Check the waveform displayed and compare it with the following.

Under
Compensated

Perfectly
CompensatedOver
Compensated

5. Use a non-metallic driver to adjust the low-frequency compensation adjustment hole on theprobe until the waveform displayed is as the “Perfectly compensated” in the figure above.

Note: It’s not necessary to compensate an active probe.

int.siglent.com

33

SDS6000L User Manual

5.5.3

LAN

Connect the LAN port to the network with a network cable with an RJ45 connector for remote control.
See the chapter “ Remote Control ” for detailed information on remotely controlling the instrument.

Follow the steps below to set LAN connection:
Utility > Menu > I/O > LAN Config
See section “LAN” for details of the configuration.

5.5.4

External Monitor and Mouse

Connect an external monitor to the HDMI output on the rear panel of the instrument using an HDMI
cable, and connect a mouse to the instrument, then it can be used as a stand-alone oscilloscope.

5.5.5

Auxiliary Output

When Mast Test is enabled, the port outputs the pass / fail signal, otherwise, it outputs the trigger
indicator. The trigger indicator can be used to measure the waveform capture rate.

See the chapter "Mask Test" for more details on the pass / fail output.

34

int.siglent.com

SDS6000L User Manual

5.5.6

Reference Input and Output

The device can use the internal 10 MHz clock or a 10 MHz clock from another instrument or source
using the 10 MHz In port as the reference. The reference clock is a 10 MHz square wave and can be
output from the 10 MHz Out port for synchronizing other instruments. To set the reference clock byfollowing the steps:
Utility > Menu > IO > Clock Source

See the "Clock Source" section for details.

5.5.7

Waveform Generator

Activate the SDS6000L-FG option to support the waveform generator function.

Perform Utility > AWG Menu to set the waveform.

See chapters “Waveform Generator” and “Bode Plot” for more relative information.

5.5.8



Logic Probe

To connect the logic probe: Insert the probe, with the correct side facing up, until you hear a“click”.



To remove the logic probe: Depress the buttons on each side of the probe, then pull out it.

See the chapter “Digital Channels” for more information.

int.siglent.com

35

SDS6000L User Manual

5.6

Power onThe SDS6000L provides two ways to power on, which are:

Auto Power-on
When the “Auto Power-on” option is enabled, once the oscilloscope is connected to the AC power
supply through the power cord, the oscilloscope boots automatically. This is useful in automated or
remote applications where physical access to the instrument is difficult or impossible.

Steps for enabling the "Auto Power-on" function:
Utility > Menu > System Setting > Auto Power On

Power on by Manual
When the " Auto Power on” option is disabled, the power button on the front panel is the only control
for the power state of the oscilloscope.

5.7

Shut down

Press the power button for one second to turn off the oscilloscope. Or follow the steps below:
Utility > Shutdown

Note: The front panel Power button does not disconnect the oscilloscope fromthe AC power supply. To disconnect the instrument from the mains, turn off
the power switch on the rear panel of the instrument. The power cord should
be unplugged from the AC outlet if the scope is not to be used for an
extended period. The standby power consumption of the 4-channel model isabout 4 W, and that of the 8-channel model is about 8 W.

36

int.siglent.com

SDS6000L User Manual

5.8

System Information

Follow the steps below to examine the software and hardware versions of the oscilloscope.
Utility > Menu > System Info

See the section "System Information" for details.

5.9

Install Options

A license is necessary to unlock a software option. See the section "Install Option" for details.

int.siglent.com

37

SDS6000L User Manual

6

Remote Control

The SDS6000L provides a LAN port and a USB Device port which can be used for remote control in
multiple ways.

Web Browser

6.1

A built-in web server provides an approach to interact with the oscilloscope by a web browser. It doesn’t
require any additional software to be installed on the computer. Set the LAN port correctly (see section“LAN” for details), input the IP address of the oscilloscope in the browser address bar, and then theuser can browse and control the oscilloscope on the web.

A.

Input the IP address of the instrument

B.

Instrument information

C. Click here to recall the instrument control interface

38

int.siglent.com

SDS6000L User Manual

Below is the instrument control interface:A.

The display and control area of the instrument’s web control interface. The display in thisarea is a copy of the instrument display. Using the mouse to operate in this area is equivalent
to directly operating the touch screen display of the instrument.

B.

Click here to perform a screenshot

C. Click here to save the waveform data as a bin file and download it to the local computer
D. Click here to save the waveform data as a CSV file and download it to the local computer

6.2

E.

Click here to download the mini-tool for converting binary files to CSVF.

Click here to perform a firmware upgrade

Other Connectivity

The SDS6000L also supports remote control of the instrument by sending SCPI commands via NIVISA, Telnet, or Socket. For more information, refer to the programming guide for this product.

int.siglent.com

39

SDS6000L User Manual

7
7.1

Screen Display
Overview

When an external monitor is connected or a web server is used, the entire screen can be manipulatedusing a mouse.

A.

Menu Bar

B.

Grid Area

C. Trigger Level Indicator
D. Cursors
E.

Channel Descriptor Boxes

F.

Trigger Delay Indicator

G. Timebase and Trigger Descriptor Boxes
H. Dialog Box

40

int.siglent.com

SDS6000L User Manual

Trigger Level Line (Vertical) and Trigger Delay Indicator (Horizontal) show the trigger position of the
waveform.

Cursors show where measurement points have been set. Move the cursors to quickly reposition themeasurement point.

Channel Descriptor boxes include analog channels (Cx), zoom traces (Zx), digital channels (D), math(Fx), memory (Mx), and reference (Ref). They are located under the grid area, showing the parametersof the corresponding traces. Clicking the boxes creates a dialog box.

Timebase and Trigger Descriptor boxes show the parameters of the timebase and trigger
respectively. Clicking the boxes creates a dialog box for the selected item.

7.2

Menu Bar

Menu bar with drop-down menus lets you access set-up dialogs and other functions.

7.3

Grid Area

The grid area displays the waveform traces. Traces can be moved by the mouse. The area is divided
into 8 (vertical) * 10 (horizontal) grids. The best display effect can be obtained by adjusting the
waveform intensity and graticule. Follow the steps below to set these parameters:

Display > Menu > Intensity ,Display > Menu > Graticule

int.siglent.com

41

SDS6000L User Manual

There are multiple indicators on the grid:
Trigger Level Indicator shows
the level where the waveform
triggers on the vertical axis.

Trigger Delay Indicator locateswhere the waveform triggers on
the horizontal axis... When thetrigger position is outside thescreen, the direction of thetriangle changes to point outside
the screen.

Channel Offset Indicator with achannel number shows the offset
position of the corresponding
channel.

The user can choose to show the axis labels or not by:
Display > Menu > Axis label Setting

Axis label = on

42

Axis label = off

int.siglent.com

SDS6000L User Manual

7.4

Channel Descriptor Box

Channel Descriptor boxes include analog channels (Cx), zoom traces (Zx), digital channels (D), math(Fx), memory (Mx), and reference (Ref). They are located under the grid area, showing the parametersof the corresponding traces. Clicking the boxes creates a dialog box. See the chapter “Vertical Setup”
for more details. Below is an example for analog channel 1:A.

Channel Index

B.

Coupling and Input Impedance

C. Vertical Scale
D. Vertical Offset
E.

Bandwidth Information

F.

Probe Attenuation Factor

Bandwidth Information:
The bandwidth information is indicated by the following icons:: 20 MHz bandwidth limit: 200 MHz bandwidth limit: Full bandwidth

Invert Indicator: shows that the current channel is inverted:: Invert has been turned on
None: Invert has been turned off

Coupling and Input Impedance:：DC coupling, 1 MΩ impedance：DC coupling, 50 Ω impedance：AC coupling, 1 MΩ impedance：AC coupling, 50 Ω impedance

int.siglent.com

43

SDS6000L User Manual：Ground

Vertical Scale: The scale of each grid in the vertical direction. For example, when the vertical scale is1.00 V/div, the full scale of the oscilloscope is 1.00 V/div * 8 div=8 V.

Vertical Offset: The offset of the channel in the vertical direction. When the vertical offset is 0, thechannel offset indicator is located in the middle of the vertical axis.

Probe Attenuation Factor: Set the probe attenuation factor to match the actual attenuation of theprobe. The oscilloscope automatically calculates the vertical scale according to the probe attenuation
factor. For example, the vertical scale of the oscilloscope is 100mV/div with 1X attenuation, and 1 V/div
if the attenuation factor is changed to 10X. When inserting a standard 10X passive probe with a probe
sense terminal, the oscilloscope will automatically set the factor to 10X.
：1:1 attenuation, suitable for direct coaxial cable connection or passive probes with 1X attenuation：10:1 attenuation, suitable for general passive probes or active probes with 10X attenuation：100:1 attenuation, suitable for some high-voltage probes： Custom attenuation factor

7.5

Timebase and Trigger Descriptor Boxes

The Timebase Descriptor box shows the current parameters of the timebase. See the chapter“Horizontal and Acquisition Setup” for more information.

A.

Trigger delay

B.

Horizontal scale（timebase）

C. # Samples
D. Sample Rate44

int.siglent.com

SDS6000L User Manual

Trigger delay: The time offset of trigger position. When the trigger delay is 0, the trigger delay indicator
is in the center of the horizontal axis of the grid area.

Horizontal scale: Timebase, the time of each grid in the horizontal direction. For example, if the scale
is 500us/div, the time of each grid is 500 us, and the full-screen time range of the oscilloscope is 500
us/div * 10 div = 5 ms.

# Samples: The number of sample points on the current screen.

Sample Rate: The current sample rate.

The Trigger Descriptor box shows the parameters of the trigger setting. See the chapter “Trigger” fordetailed information.

A.

Trigger source

B.

Trigger coupling

C. Trigger mode
D. Trigger level
E.

Trigger typeF.

Trigger slope

Trigger source

C1~C8: Analog channels

EXT: External trigger channel



EXT/5: 5x attenuation of external trigger channel

AC Line: AC mains supply



D0~D15: Digital channels

int.siglent.com

45

SDS6000L User Manual

Trigger coupling: Coupling mode of the current trigger source. It is only valid when the trigger sourceis C1~C8, EXT, or EXT/5.


DC: All the signal’s frequency components are coupled to the trigger circuit for high-frequency
bursts or where the use of AC coupling would shift the effective trigger level.



AC: The signal is capacitively coupled. DC levels are rejected. See the datasheet for details of the
cut-off frequency.



HFR: Signals are DC coupled to the trigger circuit, and a low-pass filter network attenuates highfrequencies (used for triggering on low frequencies). See the datasheet for details of the cut-off
frequency.



LFR: The signal is coupled through a capacitive high-pass filter network, DC is rejected and low
frequencies are attenuated. For stable triggering on medium to high-frequency signals. See thedatasheet for details of the cut-off frequency.

Trigger mode

Auto: The oscilloscope will sweep without a set trigger. An internal timer triggers the sweep after
a preset timeout period so that the display refreshes continuously. This is helpful when first
analyzing unknown signals. Otherwise, Auto functions the same as Normal when a trigger
condition is found.



Normal: Sweeps only if the input signal meets the trigger condition. Otherwise, it continues todisplay the last acquired waveform.

Single: Stops the acquisition and displays the last acquired waveform.



Force: Force an acquisition, regardless of whether the input signal meets the trigger conditions ornot.

Trigger level: The source voltage level or levels that mark the threshold for the trigger to fire. Trigger
levels specified in Volts normally remain unchanged when the vertical gain or offset is modified.

Trigger type: See the chapter "Trigger" for details.

46

int.siglent.com

SDS6000L User Manual

Dialog Box

7.6

The dialog box on the right side of the screen is the main area for setting the parameters of the selectedfunction.
A.

Title bar. Clicking the bar can hide thedialog box, and clicking again can open
the dialog box.

B.

Parameter setting area.

C. Scroll bar. When parameters are morethan the displayed range, the blue
scrollbar will be displayed. By sliding the
dialog area up and down, it can scroll tothe area not displayed.

To Set Parameters
The SDS6000L provides a couple of different ways to input/select parameters:
Switch: Sets parameters with two states, such as to enable ordisable a function. Click the switch region to change from one
state to the other.

List: Sets parameters with more than two
options, such as coupling mode ofchannels. Click the parameter region, andthen select the expected option from the
pop-up list.

Virtual Keypad: Sets parameters with a numerical value. Clickthe parameter region, and the parameter can be adjusted by
the mouse wheel; click the region again, then the virtual keypadappears:

int.siglent.com

47

SDS6000L User Manual

Let’s use the operation of setting the “deskew” of a channel as an example: If the expected value is 65
ns, input “65” on the virtual keypad, and then choose n as the unit “ns” to complete the operation.
On the virtual keypad, clicking the button Max , Min , and Default quickly sets the parameterto the maximum, minimum, and default values.

Hide Dialog Box
When the menu style is set to “Embedded”, the grid area will be compressed horizontally to display thecomplete waveform when the dialog box is displayed. After setting the parameters, to achieve the best
waveform display effect, click the title bar in the upper right corner to hide the dialog box. Click it again
to recall the dialog box.
When the menu style is set to “Floating”, it will directly cover part of the grid area when displayed. If
you want to observe the covered area, click the title bar in the upper right corner to hide the dialog box.
Click it again to recall the dialog box.
The dialog box can be set to be hidden automatically. When the instrument is not operated beyond theset time, the dialog box will be automatically hidden. Follow the steps below:
Display > Menu > Hide Menu .

Menu style = Floating, and the Dialog box isdisplayed

48

Dialog box is hidden

int.siglent.com

SDS6000L User Manual

7.7

Mouse Control

Waveforms, cursors, and trigger levels can be adjusted using the mouse. It can also be used to define
Zone Trigger areas as shown below:

Click-and-drag the waveform left and right tomove it on the horizontal axis

Click-and-drag the waveform up and down tomove it on the vertical axis

Click-and-drag the cursor to move it

Click-and-drag the cursor information region
to move the pair of cursors simultaneously

Click-and-drag along a diagonal line tocreate a zone or a histogram region. At the
beginning of the drawing action keep theangle close to 45°so it can be recognized as
the drawing box action

int.siglent.com

49

SDS6000L User Manual

7.8

Choosing the Language

Follow

Utility> Menu >System Setting>

Languageto choose the language. See thesection " Language " for details.

50

int.siglent.com

SDS6000L User Manual

8

Multiple Approaches to Recall Functions

The oscilloscope can recall functions through different approaches.

8.1

Menu BarIf you are familiar with common computer programs, you may first choose to access a function by thedrop-down menu from the menu bar at the top of the display.

For example, to open the trigger setup dialog box, you can follow the steps below:
Trigger > Menu
The operations can be completed with mouse clicks.

8.2

Descriptor Box

For the setup of channels, zoom traces, math, ref, memory traces, timebase, and trigger, there are
dialog boxes at the bottom of the display. For the introduction of the descriptor box, see sections"Channel Descriptor Box" and "Timebase and Trigger Descriptor Box".
For the example above, click the trigger descriptor box and thetrigger setup dialog box will be activated.

int.siglent.com

51

SDS6000L User Manual

9 Vertical Setup
9.1

Turn on/off a Channel

Click the + button and then select the expected channel to turn it on, and its channel descriptor box
and a dialog box will appear on the display. Click the channel descriptor box and then click the Off
button to disable it.

Turn on C1

Turn off C1

52

int.siglent.com

SDS6000L User Manual

9.2

Channel Setup

Click the channel descriptor box, a quick dialog will pop up. Vertical scale and offset can also be set
from this dialog box.

A.

Click the region to set the vertical scale with themouse wheel or virtual keypadB.

▲ to increase the vertical scale and ▼ todecrease

C. Check to coarsely adjust the vertical scale anduncheck to enable fine adjustment
D. Click the region to set the offset with the mousewheel or virtual keypadE.

▲ to increase the offset and ▼ to decreaseF.

Set the offset to zero

G. Copy the setting of the current channel to anotherchannelH. Quickly apply the current channel as the source ofa specified operation (Trigger, FFT, Simple
Measure, Cursor, Search, DVM, and Counter)
I.

Disable the channel

J.

Hide the trace

The vertical scale can be adjusted in coarse or fine mode. Fine mode can reach the full-scale display
as close as possible, which gets the best measurement accuracy in the vertical direction.

int.siglent.com

53

SDS6000L User Manual

Activating a channel recalls the channel dialog box, displaying more parameters:

A.

Turn the channel on/off

B.

Coupling (DC, AC, or GND)

C. Bandwidth limit (Full, 200 MHz, or 20 MHz)D. Probe settings including attenuation (1X, 10X,
100X, or custom) and probe check. If an active
probe is connected, it shows the information andsetup of the active probe
E.

Set the label text. Click to recall the label setting.
Users can customize the text and display of the
label

F.

Quickly apply a specified operation (Trigger, FFT,
Simple Measure, Cursor, Search, DVM,
Histogram, Mask Test, Counter, and AWG) to thecurrent channel

G. Impedance
H. Units for the channel
I.

DeskewJ.

Enable/disable invert

K.

Trace visible/hidden

Coupling

DC: All of the input signal frequency components are passed to the display.



AC: The signal is capacitively coupled. DC signal components are rejected. See the datasheet for
details of the cut-off frequency.AC coupling is suitable for observing AC signals with DC offset,
such as power ripple.

54

int.siglent.com

SDS6000L User Manual

GND: The channel is grounded by an internal switch. GND coupling is used to observe the zerooffset error of the analog channels or determine the source of noise in the waveform (from thesignal or from the oscilloscope itself)

Bandwidth Limit
Full bandwidth passes through signals with high-frequency components, but it also means that noisewith high-frequency components can pass through. When the frequency component of the signal isvery low, better signal-to-noise ratios (SNR) can be obtained by turning on a bandwidth limit. The
SDS6000L provides two hardware bandwidth limit options: 20 MHz and 200 MHz

Probe Settings
The SDS6000L provides 1X, 10X, 100X, and custom probe attenuation factor options. The custom
values can be between 10-6~106. The oscilloscope will automatically convert the vertical scaleaccording to the current probe attenuation factor. For example, the vertical scale of the oscilloscope
under 1X attenuation is 100 mV/div, and the vertical scale will be automatically set to 1 V/div if theprobe attenuation is changed to 10X. If a standard probe with a readout terminal is connected, theoscilloscope will automatically set the probe attenuation to match the probe.

The Probe Check option is provided to quickly check if the probe is correctly compensated. Connect
the probe as in section “Probes” and click Probe > Probe Check , then operate according to theprompt.

If an active probe is connected, click Probe to show the information and setup of the active probe.
Below is the information for an SAP1000 probe.

int.siglent.com

55

SDS6000L User Manual

For some active probes that need to be controlled from the UI, the Probe dialog provides the necessary
setup. Below is an example of a current probe, of which the Degauss, Autozero, and Coupling options
are accessible.

56

int.siglent.com

SDS6000L User Manual

Label
Users can set the label text for channels. Open the label setting dialog box to select the source,
customize the label text, and set the display. The source can be C1~C8, F1~F4, M1~M4, and
RefA~RefH. The length of the label is limited to 20 characters. The characters beyond this length will
not be displayed. When the “Display” option is set to “on”, the label will be displayed on the right sideof the channel offset indicator.

The labels are hidden

The labels are displayed

Apply toWith this setting, some common functions such as Trigger, FFT, Simple Measure, Cursor, Search,
DVM, Histogram, Mask Test, Counter, and AWG can be quickly applied to the selected channel. Oncea function is specified, it will switch directly to the function menu and automatically set that channel as
the source.

Impedance

1 MΩ: When a passive probe with high impedance is connected, the impedance must be set to 1
MΩ, otherwise the signal will not be detected.

50 Ω: Suitable for high-frequency signals transmitted through 50 Ω coaxial cables or active probes.
This can minimize the amplitude distortion caused by impedance mismatching.

Unit
Voltage unit "V" or current unit "A”. When using the current probe, the unit should be set to "A".

Deskew
Due to the skew between channels, cables, or probes, the delay of signals passing through different
measurement paths may be inconsistent. For example, two coaxial cables with a 1-inch difference inlength could introduce a skew of more than 100 ps. In some scenarios (e.g. measuring the setup/holdint.siglent.com

57

SDS6000L User Manual

time between the clock and data), it may be necessary to compensate for the skew between channels.

The method of compensation: Probe the same signal simultaneously using two channels (including thecables or probes that you intend to use for measurements) and adjust the deskew parameter of one
channel until the waveforms of the two channels observed on the screen coincide horizontally.

Before deskew

After deskew

Note: When the channel with delay compensation is the trigger source, thetrigger delay indicator will not change at any time when the deskew valuechanges.

InvertWhen invert is enabled, the waveform is rotated 180 degrees around earth potential (0 Volts).

Before invert

58

After invert

int.siglent.com

SDS6000L User Manual

Trace
When the trace is hidden, the channel waveform is no longer displayed on the screen, while theacquisition is still running in the background.

Trace visible

int.siglent.com

Trace hidden

59

SDS6000L User Manual

10 Digital Channels
10.1

Overview

SPL2016 Probe
The SPL2016 is a logic probe designed to monitor up to 16 digital signals at once. The 16 digitalchannels are separated into two groups and each group has its threshold, making it possible tosimultaneously view data from different logic families.

To avoid personal injury or damage to the logic probe and any associatedequipment, the following safety precautions should be noted.

The equipment shall be used only for the purposes specified by the manufacturer. The SPL2016
probe is used only for SIGLENT's special series of oscilloscopes. Protection mechanisms can becompromised if the way the devices connected by the SPL2016 are not used for their intended purpose.
Connect and disconnect correctly. Excessive bending can damage the cable.
Do not use equipment in humid or explosive environments.
Only used indoors. The SPL2016 is designed to be used indoors and should only be operated in aclean, dry environment.
Do not use the equipment when you suspect a problem. Do not use the SPL2016 if any parts aredamaged. Maintenance work shall be performed by maintenance personnel with appropriatequalifications.
Keep product surface clean and dry.

SDS6000L-16LA Option
60

int.siglent.com

SDS6000L User Manual

The software option adds the following functions to the oscilloscope:

Digital channel acquisition and analysis - Acquire and analyze the signals connected to thedigital logic probe, including waveform display, save, parameter measurement, etc.



Trigger on a digital channel - Trigger with the digital channel as the trigger source, isolating
events of interest.



Decode on a digital channel - Serial protocol decoding of a digital channel requires theinstallation of the serial decode option.

10.2

Enable/Disable the Digital Channels

Turning on or off the digital channels is similar to analog channels. Digital data can be stored as
waveform files. Horizontal cursors and most horizontal measurements also apply to digital waveforms.

A.

Digital channel waveform display, which shares the same grid area with the analog channels.

B.

Digital channel descriptor box

C. Dialog box
D. Digital channel indicators. Up to 16 digital channels are organized in two groups with different
thresholds: D15~D8 and D7~D0. Every channel can be turned on or off individually.
E.

Labels can be set to data, address, or custom characters.

F.

Digital bus

int.siglent.com

61

SDS6000L User Manual

Click + at the bottom of the display and select the "Digital" to turn it on. Click the digital channel
descriptor box, and select Off on the pop-up menu to turn it off. Refer to the operation in the chapter"Vertical Setup" for details.

10.3

Digital Channel Setup

Click the digital descriptor box, then the quick menu of digital channel settings pops up above thedescriptor box. In the menu height and position of digital channels can be set:A.

Upper position limit of the digital channel display area.
Use the mouse wheel or virtual keypad to set it.
Decrease the height to provide more adjustment area.

B.

▲ to increase position and ▼ to decrease the channel
location.

C. Reset the position to the default value
D. The range in height of the digital channels display
area. Use the mouse wheel or virtual keypad to set it.
If the display height covers all of the available
divisions, there will not be room to adjust the position.
E.

▲ to increase and ▼ to decrease the number ofdivisions occupied by the digital channels

F.

Reset the height to the default value

G. Turn off the digital channels
H. Open the digital dialog box on the right side

Height=8div, Position=0div

62

Height=4div, Position=1div

int.siglent.com

SDS6000L User Manual

The quick menu only covers the height range and position of the digital channels display area. Moresettings can be found in the dialog box.
A.

Turn on/off the digital channels

B.

Labels can be set to data, address, or custom
characters.

C. Logic threshold of D7~D0. The oscilloscope willautomatically set the threshold according to thespecified logic family, or the user can set thethreshold manually using the Custom option.
D. Logic threshold of D15~D8
E.

Set the channels to be displayed

F.

Deskew setting, the same as setting analog
channels

G. Bus setting

Logic Threshold Setting
The threshold level determines how the input signal is evaluated. The threshold level can be set in the
Logic Setting . The input voltage less than the threshold is recognized as a '0', and the input voltage
greater than the threshold is recognized as a ‘1’.

int.siglent.com

63

SDS6000L User Manual

The configurable logical level includes TTL, CMOS,LVCMOS 3.3 V, LVCMOS 2.5 V, and Custom.
The setting range of the custom threshold is - 10.0 V to
+ 10.0 V.

Bus Setting
Click the Digital descriptor box on the bottom to open the Digital dialog box, click Bus to open the
Digital Bus dialog box.
A.

Select Bus1 or Bus2B.

Set the data format of the digital bus (binary,decimal, unsigned decimal, andhexadecimal)

C. Set the bit width of the digital bus
D. Set the digital bus data
E.

Reset the digital bus data to the default stateF.

Enable/disable the display of the digital bus

G. Return to the previous menu

64

int.siglent.com

SDS6000L User Manual

Bus Data Setting
The bus data setting supports setting the digital channel data by bit. Click Data to open the bus datasetting dialog.

In this dialog, the mapping relationship between the bus bits and the digital channels is shown as
Bit.x[Dy]. For example, Bit.0[D15] means that bit 0 of the bus is D15. Click Bit.x[Dy] to assign thedigital channel D0-D15 to the specified bit x.

int.siglent.com

65

SDS6000L User Manual

11 Horizontal and Acquisition Setup
11.1

Timebase Setup

The timebase setup is used to adjust the scale and offset of the X (horizontal) axis. This setting applies
to all analog, and digital channels as well as all math traces except FFT.

Click the timebase descriptor box and the quick menu of the timebase settings will pop up. In the menu
timebase (horizontal scale), delay and other parameters can be set.
A.

Set the horizontal scale (timebase) by thevirtual keypadB.

▲ to increase and ▼ to decrease thehorizontal scale

C. Set the trigger delay by the virtual keypad
D. ▲ to increase and ▼ to decrease the trigger
delay
E.

Set the trigger delay to zeroF.

Set the trigger delay to the left part of thescreenG. Set the trigger delay to the right part of thescreenH. Open the Acquire dialog box

66

int.siglent.com

SDS6000L User Manual

11.2
11.2.1

Acquisition Setup
Overview

Click Acquire Menu on the quick menu of the timebase settings, or click the menu bar
Acquire > Menu to recall the Acquire dialog box on the right side.
A.

Select the interpolation mode

B.

Select the Acq mode

C. Select the acquisition mode (Normal / Peak /Average / ERES). See the section "Acquisition"
for details.
D. Select the Memory Management mode (Auto,
Fixed Sample Rate, and Fixed Memory). See
the section "Memory Management" for details.
E.

Select the maximum memory depth

F.

Enter the sequence menu. See the section“Sequence” for details

G. ESR mode on/off. See the section “ESR” fordetails
H. XY mode on/off

Interpolation -- At small timebase settings, the number of original points on the screen may be lessthan the number of display pixels in the grid area, so interpolation is necessary to display a continuous
waveform. For example, at 1 ns/div timebase and 5 GSa/s sample rate, the number of original points
is 50, but the grid area includes 1250 horizontal pixels. In this case, the oscilloscope needs tointerpolate the original points by 25.

int.siglent.com

67

SDS6000L User Manual

X: Linear interpolation, the simplest way of interpolation, connects two original points with astraight line.

Sinc: Sin(x)/x interpolation, the original point is interpolated according to the Nyquist
reconstruction formula, which has a good time-domain recovery effect for a sine wave. But for step
signals / fast rise times, it will introduce false overshoot due to the Gibbs phenomenon.

X Interpolation

Sinc Interpolation

Acq mode: "Fast" is the default setting. The SDS6000L provides a very high waveform update rate infast mode. "Slow" mode will slow down the waveform update on purpose.

Memory Depth: The memory depth that can be supported. According to the formula "acquisition time
= sample points x sample interval", setting a larger memory depth can achieve a higher sample rate
for a given time base, but more samples require more processing time, degrading the waveform update
rate.

The memory depth supported by the SDS6000L is listed in the following table:

68

Single Channel Mode

Dual-Channel Mode

Four-Channel Mode

2.5 k

2.5 k

2.5 k

5k

5k

5k

25k

25k

25k

50k

50k

50k

250k

250k

250k

500k

500k

500k

2.5M

2.5M

2.5M

5M

5M

5M
int.siglent.com

SDS6000L User Manual

12.5M

12.5M

12.5M

25M

25M

25M

50M

50M

50M

125M

125M

125M

250M

250M

500M

Channels 1, 2, 3, and 4 (C1~C4) are in group 1 and 5, 6, 7, and 8 (C5~C8) are in group 2Single-channel mode: In one group, only one channel is on and in the other group, one channel
or less is on.

Dual-channel mode: In one group, two channels are turned on and in the other group, two channels
or less are turned on.

Four-channel mode: In any group, three or more channels are turned on.

11.2.2Acquisition

The acquisition mode is used to determine how to acquire and process the signal.

Normal: The oscilloscope samples the signal with equal time intervals. For most waveforms, the best
display effect can be obtained using this mode.

Peak: Peak detect mode. The oscilloscope acquires the maximum and minimum values of the signal
within the sample interval so the peak (maximum-minimum) in the interval is obtained. This mode iseffective to observe occasional narrow pulses or spurs with a low sample rate, but the noise displayed
is larger. In peak mode, the oscilloscope will display all pulses with a pulse width longer than 200 ps.

In the following example, a narrow pulse sequence with a pulse width of 3.4 ns and a period of 200 Hz
is sampled at a 5 MSa/s sample rate in normal mode and peak mode. As we can see, because thesample interval (200 ns) is much larger than the pulse width (3.4 ns), it is difficult to capture the narrowpulses in normal mode, but peak mode can ensure that each pulse is captured.

int.siglent.com

69

SDS6000L User Manual

Normal mode

Peak mode

Average: The oscilloscope accumulates multiple waveform frames and calculates the average as theresult. If a stable trigger is available, the resulting average has a random noise component lower thanthat of a single-shot record. The more frames that are accumulated, the lower the noise is. For the
SDS6000L, average processing is implemented by the hardware engine, so it can still maintain a highwaveform update rate when the acquisition mode is set to average.

Normal mode

Average mode（32）Note: Average acquisition is only valid for periodic signals, and it is important
to ensure that the waveform is triggering stably when using average mode.

ERES: In this mode, the oscilloscope digitally filters the sampling points, which can reduce thebroadband random noise on the input signal and improve the signal-to-noise ratio, to improve theeffective resolution (ENOB) of the oscilloscope. The ERES processing of the SDS6000L is realized by
the hardware engine, so it can still maintain a high waveform refresh rate when ERES is turned on.

70

int.siglent.com

SDS6000L User Manual

Normal mode

ERES mode (3-bit)

ERES acquisition does not require the signal to be periodic, nor does it require stable triggering, but
due to the digital filtering, the system bandwidth of the oscilloscope will degrade in ERES mode. The
higher the enhanced bits, the lower the bandwidth. The following table shows the relationship between
ERES bits and bandwidth:

Enhanced Bits

Equivalent Hi-Res
Enhanced Resolution-3dB bandwidth0.5

10.25*Sample rate

12

0.115*Sample rate

1.5

30.055*Sample rate2

40.028*Sample rate

2.5

50.014*Sample rate

3

6

0.007*Sample rate

3.5

70.0035*Sample rate

480.0017*Sample rate

int.siglent.com

71

SDS6000L User Manual

11.2.3

Memory Management

Memory Management controls how the instrument stores the acquired samples.

Auto: The default acquisition setting. After setting the maximum memory in the Auto mode, theoscilloscope automatically adjusts the sample rate and memory depth according to the time base. The
adjustment principle is to follow the formula:Sample rate = number of samples/time of acquisition
Where the time of acquisition is the time corresponding to the full display (i.e. 10 horizontal divisions).

Note: The maximum memory here is the upper limit of the memory space
allocated by the oscilloscope. The actual sample points are related to thecurrent time base and may be less than memory depth. The actual sample
points information can be obtained in the timebase descriptor box (see thesection "Timebase and Trigger" for details).

Fixed Sample Rate: The sample rate is fixed as set, and the oscilloscope automatically adjusts
memory depth according to the time base. The time of acquisition is the time corresponding to the full
display (i.e. 10 horizontal divisions).

Fixed Memory: The memory depth is fixed as set, and the oscilloscope automatically adjusts the
sample rate according to the time base. With a small time scale (such as 1 ns/div) the memory depth
is beyond the time of the full display, so it needs to zoom out the acquired frame for viewing thecomplete frame in Stop mode.

Note: When the status is Run, the scope only processes the data on thedisplay, which means measure, math, decode, search, etc. only analyzes thedata on the display. This is identical to the "Auto" memory management
mode.

72

int.siglent.com

SDS6000L User Manual

Roll Mode

11.2.4

Click Acquire > Roll to enter roll mode. In this mode, the waveform moves across the screen
from right to left, similar to a strip chart recorder. The horizontal delay control of the waveform will bedisabled when roll mode is active. It only operates at timebase values of 50 ms/div and above.
If you would like to stop the display in Roll mode, click Acquire > Run/Stop . To clear the display
and restart an acquisition in Roll mode, perform Acquire > Run/Stop again.

Note: Zoom is only supported after Stop in Roll mode.

Sequence

11.2.5Click Acquire > Sequence to recall the sequence dialog box.
A.

Turn on or off sequenceB.

Set the segment

Sequence mode is a fast acquisition mode, which divides the memory depth into multiple segments(up to 80,000), each of which stores a single shot. In sequence mode, the oscilloscope only acquires
and stores data without processing and displaying it until the specified segments are acquired. As aresult, the dead time between trigger events is minimized, thus greatly improving the waveform update
rate. If sequence mode is enabled, the display will not update until all of the sequences have beenacquired. The SDS6000L can achieve a minimum 1.3 us trigger interval in Sequence mode,
corresponding to a waveform update rate of 750,000 wfm/s.

After the acquisition is finished, the oscilloscope will map all the segments together to the screen. If
you need to view and analyze each frame separately, history mode will help (see the section "History"
for details). History mode provides timestamp labels for each segment.

int.siglent.com

73

SDS6000L User Manual

In addition to minimizing the dead time, another advantage of Sequence mode is that it can capture
and record rare events over long periods. The oscilloscope can capture multiple events that satisfy thetrigger conditions, ignoring the periods of no interest between adjacent events, thus maximizing theuse of waveform memory. You can use the full accuracy of the acquisition timebase to measureselected segments.
Sequence N-1

Sequence N

Sequence N+1

Stored Waveform

Example:Input a pulse sequence with a period of 50 ms to C1. The rise time of the pulse is 2 ns, the fall time is
100 ns, the pulse width is 108 ns, and amplitude is 1.6 Vpp. Perform Acquire > Auto Setup .
Set the coupling mode of C1 to DC50Ω, and vertical scale to 500 mV/div, vertical offset to 0. Set thetrigger level to 0.

74

int.siglent.com

SDS6000L User Manual

In normal mode, 5 pulses can be obtained on the screen with the sample rate of 1.25 GSa/s
at the maximum memory depth.

Set the trigger mode to "Single" and the timebase to 50 ns/div. Turn on the Sequence mode, and set
the segments to maximum (46,378 in this example, up to 80,000 depending on the number of samples
at the current time base). Wait patiently until the acquisition completes, then all the waveforms
satisfying the trigger conditions are displayed on the screen.

In Sequence mode, there is no waveform displayed on the screen until the acquisition iscompleted. During acquisition, there is a counter on the screen indicating the number ofsegments that have been acquired.

int.siglent.com

75

SDS6000L User Manual

In the example, 53969 pulses can be obtained with the sample rate of 5 GSa/s at the
maximum memory depth.

11.2.6

ESR

Each channel of the SIGLENT SDS6000L Oscilloscope employs a 5 GSa/s analog-to-digital-converter(ADC). After each ADC, a 2x interpolation process is applied that provides a 10 GSa/s ESR option for
the users. The data flow and corresponding spectrum are shown in the figure below:

A/D
2X

Analog5 GSa/s

10 GSa/s

1st Nyquist Zone

Fs = 5 GSa/s

1st Nyquist Zone

Fs = 10 GSa/s

ESR data flow in the SDS6000L

76

int.siglent.com

SDS6000L User Manual

The 2x interpolation implemented is a classic sinx/x up-sampling processed by a high-speed Field
Programmable Gate Array (FPGA) in real-time. The following figure shows the frequency response of
the Finite Impulse Response (FIR) digital filter used in the interpolation. From the frequency response,
we can see it has no attenuation at fmax = 2 GHz, and ~70 dB attenuation at 5 GHz – fmax = 3 GHz,
which provides enough alias suppression for signals in the operating bandwidth. The aliasing
associated with signals above 2 GHz is minimized by the fast roll-off of the analog front end.

Higher measurement precision
The ADC sample rate to bandwidth ratio in the SDS6000L is 2.5:1. That means for a 2 GHz input there
are 2.5 samples/period. Without any extra interpolation, the measurements on vertical waveform
elements such as Maximum, Minimum, and Peak-peak will introduce considerable error at highfrequency input. The figure below shows an example in which a 500 MHz signal is sampled at 5 GSa/s:

Error on vertical measurement introduced by insufficient sampling
int.siglent.com

77

SDS6000L User Manual

Interpolation can significantly reduce this kind of error. Below is the same signal with an identicalsample rate but followed by the 2x interpolation technique. From this figure, we can see points inserted
between the raw data in the top/base area are closer to the real top/base of the analog input, which
reduces the error.

Error on vertical measurement after 2x interpolation

Higher ERES bandwidth
High-resolution acquisition in the SDS6000L enhances the vertical resolution but doing so results in adecrease in the available bandwidth. The higher the resolution, the lower the bandwidth. The bandwidth
as a function of resolution and sample rate is shown in the table below:

Table: Bandwidth specifications of the SDS6000L in ERES acquisition mode
0.5-bit: 0.25*Sample rate, up to the analog bandwidth
1-bit: 0.115*Sample rate, up to 1.15 GHz, limited by the analog bandwidth
1.5-bit: 0.055*Sample rate, up to 550 MHz, limited by the analog bandwidth
Bandwidth in
ERES mode(typical)

2-bit: 0.028*Sample rate, up to 280 MHz
2.5-bit: 0.014*Sample rate, up to 140 MHz
3-bit: 0.007*Sample rate, up to 70 MHz
3.5-bit: 0.0035*Sample rate, up to 35 MHz
4-bit: 0.0017*Sample rate, up to 17 MHz

78

int.siglent.com

SDS6000L User ManualWith the ESR enabled, in ERES acquisition mode, the bandwidth will be twice as scenarios without
ESR:

Table: ERES Bandwidth comparison between ESR disabled and enabled
Enhanced

Bandwidth with ESR = OFF

Bandwidth with ESR = ON

Resolution(5 GSa/s)(10 GSa/s)

0.5-bit1.25 GHz

2 GHz

1-bit

575 MHz1.15 GHz

1.5-bit275 MHz

550 MHz

2-bit

140 MHz280 MHz

2.5-bit70 MHz140 MHz

3-bit

35 MHz70 MHz3.5-bit17.5 MHz

35 MHz4-bit

8.5 MHz17 MHzLimitations
The ESR algorithm has its limitations. It cannot enhance the hardware performance of an ADC andintroduces extra overshoot/undershoot and ringing on fast edges.

With this understanding, the user can choose to enable ESR or not, according to the actual application.

int.siglent.com

79

SDS6000L User Manual

11.3

History

Click Analysis > History to recall the history dialog box.
A.

Turn on or off history mode

B.

Specify the frame index

C. Set the playing mode
D. Play backward, pause, and play forward
E.

Turn on or off the list

F.

List time type

G. Set the time interval between two frames
when playing automatically
H. Enable the “Stop on Search Event”
function, which allows navigating search
events across history frames.
I.

List displays the frame index and timelabel for each frame.

The oscilloscope automatically stores acquired frames. It can store up to 80,000 frames but the number
may vary due to the memory depth and timebase settings. Turn on history mode, then the stored
frames can be recalled and measured.

Continue with the example in the section above. In Sequence mode, all waveforms that satisfy thetrigger conditions are mapped to the display. If you need to observe a single frame, you can use historymode.

To enable history mode, click the Frame No . area twice, then the virtual numeric keypad pops up.
Input the number "5412" to specify the 5412th segment (frame).

80

int.siglent.com

SDS6000L User Manual

Observing the 5412th frame captured by Sequence in history mode

Click the List area and turn on the list. The time label corresponding to the 5412th waveform isdisplayed. The time resolution is microseconds. Time label types include AcqTime or Delta T ,AcqTime corresponds to the absolute time of the frame, synchronized with the real-time clock of theoscilloscope; Delta T is the acquisition time interval between adjacent two frames, it is shown as 50ms
in the following diagram, which is consistent with the period of the actual waveform.

Acq Time label

int.siglent.com

Delta T label

81

SDS6000L User Manual

In addition to manually specifying a frame, history mode supports auto-play:
Click thesoftkey to replay the waveform from the current frame to the first.

Click thesoftkey to stop replay.

Click thesoftkey to replay the waveform from the current frame to the last.

Click the Interval Time area to control the speed of automatic play. In the process of automatic play,
the list will automatically scroll to the current frame.

Note: When a measurement is enabled in history mode, if the Interval Time isset to small (e.g. in us) the speed of automatic play may not meet the setting,because measurements may take longer for each frame.

82

int.siglent.com

SDS6000L User Manual

12 Zoom
The SDS6000L supports waveform zoom in the horizontal and vertical directions. Perform
Acquire > Zoom to turn on Zoom, or click + at the bottom to add a zoom trace.
When the Zoom function is on, the waveform area is divided into upper and lower parts. The area of
about 1/3 height above is the main window, and the area of about 2/3 height below is the zoom window.
Click a window to activate it.

The area not covered by the gray background in the main waveform area is the range to be zoomedin (zoom area). Click the descriptor box of the zoom trace to recall the quick dialogs for setting thevertical and horizontal parameters of the zoom window.

Horizontal and vertical setting

int.siglent.com

83SDS6000L User Manual

Below is an example of changing the setting in the horizontal direction by different actions. The actions
for changing the setting in the vertical direction are similar.

Adjust the horizontal position of the zoom area by dragging left and right in the zoom area of themain window or waveform in the Zoom window.
Adjust the vertical position of the zoom area by dragging up and down in the zoom area of themain window or waveform in the Zoom window.
Adjust the horizontal position of the waveformby dragging the gray area of the main window
left and right.
Adjust the vertical position of the waveform by
dragging the gray area of the main window up
and down.

Note: Zoom traces (Z1~Z8) can be used as the source of Measurement,
Math, Mask Test, and Histogram.

84

int.siglent.com

SDS6000L User Manual

13 Trigger
13.1

Overview

The oscilloscope only acquires waveforms of interest (i.e. the ones that satisfy the trigger condition)
and aligns all trigger events at the trigger position to form a stable waveform display. The trigger is oneof the most important features of an oscilloscope since we can only analyze a signal that we can trigger
reliably and stably.

The trigger position is movable on the display. The following diagram shows the structure of theacquisition memory. The acquisition memory is divided into pre-trigger and post-trigger buffers and theboundary between them is the trigger position. Before the trigger event arrives, the oscilloscope fills
the pre-trigger buffer first, and then continuously updates it in FIFO mode until the trigger event arrives.
After the trigger event, the data fills the post-trigger buffer. When the post-trigger buffer is full, an
acquisition is completed.

Trigger Event

Pre-trigger Buffer

Post-trigger Buffer

Acquisition Memory

Below is the definition of the states in the process of filling the acquisition memory:
Arm: The pre-trigger buffer is not full, and the oscilloscope does not respond to any trigger events.
Ready: The pre-trigger buffer is full, and the oscilloscope is waiting for the trigger event.
Trig’d: A trigger event is detected and the oscilloscope starts to fill the post-trigger buffer.

Trigger settings should be based on the features of the input signal. For example, a sine wave with arepeatable period can be triggered on the rising edge; for capturing hazards in a combinational logic
circuit, the pulse trigger can be set. You need to have some knowledge of the signal-under-test to
quickly capture the desired waveform.

int.siglent.com

85

SDS6000L User Manual

13.2

Trigger Setup

Click the trigger descriptor box to display the quick menu of trigger settings. The trigger setup dialog
box is displayed on the right side of the screen.
A.

Click the level region and roll the mouse wheel toadjust the trigger level. Click the region again andthe virtual keypad will be available

B.

▲ to increase and ▼ to decrease the trigger level

C. Set the trigger level to the vertical center of the
waveform automatically
D. Set the trigger mode to "Auto"E.

Set the trigger mode to "Single"F.

Set the trigger mode to "Normal"

Quick menu

G. Perform a force trigger

Trigger setup dialog box:
A.

Select trigger type

B.

Select the trigger source

C. Select the trigger slope (When the trigger type is"Edge", "Slope" and other specific types)
D. Set holdoff (None/Time/Events)
E.

Set trigger coupling mode (DC/AC/LF Reject/HF
Reject)F.

Enable/disable Noise Rejection. When Noise
Reject is on, the trigger hysteresis is increased,so the noise immunity of the trigger circuit isbetter. As a compromise, the trigger sensitivity
degradesG. Set the Zone trigger

86

int.siglent.com

SDS6000L User Manual

Trigger Related Label

Trigger level Indicator

Horizontal 0 position Indicator

Horizontal 0 position(out of screen) Indicator

13.3

Trigger Level

Both analog and digital triggers must have a correct trigger level value. The oscilloscope judges
whether a waveform satisfies the trigger condition when it crosses the trigger level. If it does, thecrossing time is the trigger position. In the following figure, the trigger condition is set as a rising edge.
When the signal with a positive slope crosses the trigger level, the trigger condition is satisfied and thetime point the signal crosses the level is the trigger position.

In some special triggers, the system will automatically set the trigger level, such as using AC Line as
the trigger source.

int.siglent.com

87SDS6000L User Manual

13.4

Trigger Mode

The trigger mode determines how the oscilloscope acquires waveforms.

Auto: An internal timer triggers the sweep after a preset timeout period if no trigger has been foundso that the oscilloscope continuously updates the display whether a trigger happens or not. Automode is suitable for unknown signals or DC signals.

Note: In Auto mode, if the signal satisfies the trigger conditions but cannot
trigger the oscilloscope stably, it may be that interval between two trigger
events exceeds the timeout period. Try Normal mode in this case.

Normal: Triggers and acquisitions only occur when the trigger conditions are met. Otherwise, theoscilloscope holds the last waveform on the display and waits for the next trigger. Normal mode issuitable for acquiring:

Only events specified by the trigger settings

Rare events

Single: Captures and displays a single frame that satisfies the trigger conditions, and then stops. The
following trigger events are ignored until the Single acquisition is restarted. Single-mode is suitable for:One-shot events or periodic signals, such as power-on\off waveforms on a power railRare events

Force: The oscilloscope triggers regardless of whether the input signal meets the trigger conditions or
not.

88

int.siglent.com

SDS6000L User Manual

13.5

Trigger Type13.5.1

Overview

The trigger modes of the SDS6000L are digital designs. Compared with analog trigger circuits, digitaltriggers can not only greatly optimize trigger precision and trigger jitter, but also support multiple trigger
types and complex trigger conditions.
Edge -- Trigger on a rising edge, falling edge or both
Slope -- Trigger when an edge crosses two thresholds that lie inside or outside aselected time range
Pulse -- Trigger at the end of a pulse when the pulse width lies inside or outside aselected time range
Video -- TV trigger on falling edge

Window -- Trigger when the signal leaves the widow region.
Interval -- Trigger on the second edge when the time between the edges is inside oroutside a selected time range
Dropout -- Trigger when the signal disappears for longer than the Dropout value.
Runt -- Trigger when a pulse crosses the 1st threshold but not the 2nd before recrossing the 1st threshold again
Pattern -- Trigger when pattern condition transitions from false to true. All inputs set
to DC coupling
Serial --Trigger on specified condition in a serial bus. See the chapter "Serial Trigger
and Decode" for details.
Qualified -- Trigger with edge trigger setting only after the qualifying condition issatisfied
Nth Edge –Trigger on the Nth edge of a burst that occurs after a specified idle time
Delay --Trigger when the delay time between source A and source B meets the limit
condition
Setup/Hold --Trigger when the setup time or hold time meets the limit condition

int.siglent.com

89SDS6000L User Manual

13.5.2

Edge Trigger

Edge trigger distinguishes the trigger points by seeking the specified edge (rising, falling, alternating)and trigger level. The trigger source and slope can be set in the trigger dialog box.

Click the Source area to select the trigger source, and click the Slope area to select rising, falling,or alternating.
Rising -- Only trigger on the rising edge
Falling -- Only trigger on the falling edge
Alternating -- Trigger on both the rising and falling edge

Holdoff, coupling, and noise reject can be set in edge trigger, see the sections "Holdoff", "Trigger
Coupling" and "Noise Reject" for details.

Note: Perform Auto Setup , and the oscilloscope will set the trigger type to
Edge.

13.5.3Slope Trigger

The slope trigger looks for a rising or falling transition from one level to another level in the specified
time range. For example, positive slope time is defined as the time difference between the two crossing
points of trigger level lines A and B with the positive edge as shown in the figure below.

90

int.siglent.com

SDS6000L User Manual

Click the Source area to select the trigger source, and click the Slope area to select rising or
falling.
Rising -- Only trigger on the positive slope
Falling -- Only trigger on the negative slope

Adjust the Upper/Lower Level
The slope trigger requires upper and lower trigger
levels. When the trigger type is slope trigger, click
the trigger descriptor box, the pop-up quick menu
will show two levels.

The upper/lower level can be set in the following
way:

Click the Level Upper area in the quick menu to select the upper level, and then set the level value
by the virtual keypad or the mouse wheel. To set the lower level is similar.

The lower level should always be less than or equal to the upper level. In the trigger descriptor box,
the lower level is displayed.

int.siglent.com

91

SDS6000L User Manual

Set Limit Range
Click theLimit Range

area in the trigger dialog box to select the time condition and set thecorresponding time in the Upper Value / Lower Value area.
Less than a time value（≤） -- Trigger when the positive or negative slope time of the input signal islower than the specified time value.
Great than a time value（≥） -- Trigger when the positive or negative slope time of the input signal
is greater than the specified time value.
Within a range of time values （[--,--]） -- Trigger when the positive or negative slope time of the
input signal is greater than the specified lower limit of time and lower than the specified upper limit oftime value.
Outside a range of time value（--][--） -- Trigger when the positive or negative slope time of the input
signal is greater than the specified upper limit of time and lower than the specified lower limit of timevalue.

Holdoff, coupling, and noise reject can be set in slope trigger, see the sections " Holdoff ", " Trigger
Coupling " and "Noise Reject" for details.

13.5.4

Pulse Trigger

Trigger on a positive or negative pulse with a specified width. Trigger source, polarity (positive,
negative), limit range, and time value can be set in the trigger dialog box.

92

int.siglent.com

SDS6000L User ManualLess than a time value（≤）-- Trigger when the positive or negative pulse time of the input signal islower than the specified time value. Below is an example of a trigger condition set to a positive pulse
width < 100ns

Great than a time value（≥）-- Trigger when the positive or negative pulse time of the input signal isgreater than the specified time value. Below is an example of a trigger condition set to a positive pulse
width > 100ns

Within a range of time values （[--,--]）-- Trigger when the positive or negative pulse time of the input
signal is greater than the specified lower limit of time and lower than the specified upper limit of thetime value. Below is an example of a trigger condition set to 100 ns < positive pulse width < 300 ns.

Outside a range of time value（--][--）-- trigger when the positive or negative pulse time of the input
signal is greater than the specified upper limit of time and lower than the specified lower limit of thetime value.

Holdoff, coupling, and noise reject can be set in pulse trigger, see the sections "Holdoff", "Trigger
Coupling" and "Noise Reject" for details.

int.siglent.com

93

SDS6000L User Manual

Video Trigger

13.5.5

Video trigger can be used to capture the complicated waveforms of most standard analog video signals.
The trigger circuitry detects the vertical and horizontal interval of the waveform and produces a trigger
based on the video trigger settings you have selected. The SDS6000L supports standard video signals
for NTSC (National Television Standards Committee), PAL (Phase Alternating Line), HDTV (HighDefinition Television), and a custom video signal trigger.

Source, standard, and synchronization mode can be set in the video trigger dialog box. When thesynchronization mode is "Select", line and field can be specified.

Click theStandard

and select the video standard. The SDS6000L supports the following video

standards:
TV Standard

Scan TypeSync Pulse

NTSC

Interlaced

Bi-level

PAL

Interlaced

Bi-level

HDTV 720P/50

Progressive

Tri-level

HDTV 720P/60

Progressive

Tri-level

HDTV 1080P/50

Progressive

Tri-level

HDTV 1080P/60

Progressive

Tri-level

HDTV 1080i/50

Interlaced

Tri-level

HDTV 1080i/60

Interlaced

Tri-level

Custom94

int.siglent.com

SDS6000L User Manual

The table below shows the parameters of the custom video trigger:
Frame Rate

25Hz, 30 Hz, 50 Hz,60 Hz

Of Lines

300 ~ 2000

Of Fields

1,2,4,8Interlace

1:1, 2:1,4:1,8:1

Trigger Position

Line

Field

(line value)/1(1:1)

1(line value)/2 (2:1)

1,2, 3, 4,5,6,7,8(line value)/4(4:1)

1,2, 3, 4,5,6,7,8(line value)/8(8:1)

1,2, 3, 4,5,6,7,8

Line value: The number of lines set in the Of Lines (300 ~ 2000).

In the custom video trigger type, the corresponding "Of Fields" varies with the selection of the “Interlace”
ratio. Therefore, the number of fields selected and the number of lines corresponding to each field canalso be varied. If the "Of Lines" is set to 800, the correct relationship between them is as follows:
Of Lines

Interlace

Of Fields

Trigger Line

Trigger Field

800

1:1

1

800

1

800

2:1

1/2/4/8400

1/1~2/1~4/1~8

800

4:1

1/2/4/8200

1/1~2/1~4/1~8

800

8:1

1/2/4/8100

1/1~2/1~4/1~8

Set the video trigger for a video signal
Click Sync for trigger mode selection, video trigger mode has "Any" and "Select" options. In "Any"mode, the video signal can be triggered on any line that meets the conditions. In the "Select" mode,the signal can be triggered by a specified field and a specified line.

int.siglent.com

95

SDS6000L User ManualFor progressive scanning signals (e.g. 720p/50, 720p/60, 1080p/50, 1080p/60), only the specified line
can be selected to trigger when the synchronization mode is “Select”.
For interlaced scanning signals (such as NTSC, PAL, 1080i/50, 1080i/60, and Custom), when thesynchronization mode is "Select", the specified line and the specified field can be selected to trigger.

The following table shows the corresponding relations between line and field for all video standards(except for Custom)Standard

Field 1

Field 2NTSC

1 to 263

1 to 262

PAL

1 to 313

1 to 312

HDTV 720P/50 , 720P/60

1 to 750

HDTV 1080P/50 , 1080P/60

1 to 1125

HDTV 1080i/50 , 1080i/60

1 to 563

1 to 562

To gain familiarization with the video trigger, try these two examples:

Trigger on a specific line of video (NTSC standard)

Use “Custom” to trigger video signals

Trigger on a Specific Line of Video
Video trigger requires that any analog channel can be used as the trigger source with a synchronization
amplitude greater than 1/2 grid. The example below sets to trigger on Field 1, Line 22 using the NTSC
video standard.
1.

Open the trigger menu.

2.

In the trigger menu, click Type , and select "Video".

3.

Click the Source and select CH1 as the trigger source.

4.

Click the Standard and select the "NTSC".

5.

Click the Sync and select the "Select" to make the Field and Line optional, then select "1"
in the "Field", and set the "Line" to "22" by using the mouse wheel or the virtual keypad.

96

int.siglent.com

SDS6000L User Manual

Trigger on a Specific Line of Video (NTSC)

Use "Custom" to Trigger Video Signals
Custom video trigger supports video signals with frame rates of 25, 30, 50, and 60 Hz respectively,
and the specified row is within the range of 300 to 2000. The following describes how to trigger a"Custom" video signal.
1. Open the trigger menu.
2. In the trigger menu, click the Type , and select "Video".
3. Click the Source and select CH1 as the trigger source.
4. Click the Standard and select the "Custom".
5. Click the Custom Setting to open the custom setting menu and click the Interlace toselect the required interlace ratio (assuming that the interlace ratio is 8:1). Then set the frame
rate and select the number of lines and the number of fields.
6. Click the Sync to select the synchronization mode for the input signal:a)Select the "Any" mode, and the signal can be triggered on any line that meets the trigger
condition.

b)

Select the "Select" mode, then set the specified line and the specified field to trigger thesignal. Assuming that the "Field" is set to 8, you can select any field from 1 to 8, andeach field can choose any line from 1 to 100.

int.siglent.com

97

SDS6000L User Manual

13.5.6

Window Trigger

The window trigger is similar to the edge trigger, except that it provides an upper and a lower trigger
level. The instrument triggers when the input signal passes through the upper level or the lower level.

There are two kinds of window types: Absolute and Relative. They have different trigger-leveladjustment methods. Under the Absolute window type, the lower and the upper trigger levels can be
adjusted separately. The relative window provides adjustment for the Center value to set the window
center and the Delta value to set the window range. In this mode, the lower and the upper trigger levels
always move together.



If the lower and the upper trigger levels are both within the waveform amplitude range, theoscilloscope will trigger on both rising and falling edges.



If the upper trigger level is within the waveform amplitude range while the lower trigger levelis out of the waveform amplitude range, the oscilloscope will trigger on the rising edge only.



If the lower trigger level is within the waveform amplitude range while the upper trigger levelis out of the waveform amplitude range, the oscilloscope will trigger on the falling edge only.

To set window trigger via the Absolute window type
Refer to "Adjust Upper/Lower Level" in the section "Slope Trigger".

98

int.siglent.com

SDS6000L User Manual

To set window trigger via the Relative window type
When the window trigger type is set to "Relative",click the trigger descriptor box. The pop-up
menu will show two user-defined parameters:“Level +/-Delta" and "Level Center".

The above two parameters can be set in thefollowing way:

Select the parameter in the Level +/-Delta area of the quick menu, then set the parameter value by
the virtual keypad or the mouse wheel. Setting the center level is similar.

Note: "Level +/-Delta” represents half of the actual window area. For example,
when the value is 200mV, it represents a range of ±200 mV, which is a 400mV
window.

Holdoff, coupling, and noise reject can be set in the window trigger, see the sections "Holdoff", "Trigger
Coupling" and "Noise Reject" for details.

13.5.7

Interval Trigger

Trigger when the time difference between the neighboring rising or falling edges meets the time limit
condition.

When the trigger condition is set as an interval between two neighboring rising edges and it is less than
the set time value, the trigger diagram is as follows:

int.siglent.com

99

SDS6000L User Manual

Trigger source, slope (rising, falling), limit range, and time value can be set in the trigger dialog box.
Holdoff, coupling, and noise reject can be set in interval trigger, see the sections "Holdoff", "Trigger
Coupling" and "Noise Reject" for details.

13.5.8

Dropout Trigger

Dropout trigger includes two types: Edge and State.

Edge
Trigger when the time interval (△T) from when the rising edge (or falling edge) of the input signal
passes through the trigger level to when the neighboring rising edge (or falling edge) passes throughthe trigger level is greater than the set time, as shown in the figure below:

State
Trigger when the time interval (△T) from when the rising edge (or falling edge) of the input signal
passes through the trigger level to when the neighboring falling edge (or rising edge) passes throughthe trigger level is greater than the set time, as shown in the figure below:

Trigger source, slope (rising, falling), dropout type, and time value can be set in the trigger dialog box.
100

int.siglent.com

SDS6000L User Manual

Holdoff, coupling, and noise reject can be set in the dropout trigger, see the sections "Holdoff", "Trigger
Coupling" and "Noise Reject" for details.

13.5.9Runt Trigger

Runt trigger looks for pulses that cross one threshold but not another as shown in the figure below:

A positive runt pulse across through the low level but not the high level.

A negative runt pulse across through the high level but not the low level.

Holdoff, coupling, and noise reject can be set in the dropout trigger, see the sections "Holdoff", "Trigger
Coupling" and "Noise Reject" for details.

13.5.10

Pattern Trigger

The Pattern trigger identifies a trigger condition by looking for a specified pattern. The SDS6000L
provides 4 patterns: logical AND, OR, NAND and NOR combination of the channels. Each channel canbe set to "Low", "High" or "Don't Care". If all channels are set to "Don’t Care", the oscilloscope will not
trigger. The result after the logical operation is triggered from true to false, that is, triggers on the fallingedge of the combined waveform.

int.siglent.com

101

SDS6000L User Manual

Logic (AND, OR, NAND, NOR), source, limit range, and time value can be set in the trigger dialog box.

Source Setting
Click theSource Settingarea to recall the following dialog box and set up for each channel

separately. Each channel can be set to "Low", "High" or "Don't Care". The threshold can be determinedby setting the Level Value .
When digital channels are turned on, the logic state of the digital channel can also be set in the source
setting dialog box.

102

int.siglent.com

SDS6000L User Manual

The logical setting of analog channels

The logical setting of digital channels

Limit Range
This setting is particularly useful to filter the hazard signals of combinational logic.

Holdoff can be set in pattern trigger, see the section "Holdoff" for details.

13.5.11

Qualified Trigger

The qualified trigger is an edge trigger after a user-defined qualified condition is satisfied. So a qualified
trigger has two sources; one is the source of the edge trigger, and the other is the qualified source.
The qualified types include “State”, “State with Delay”, “Edge”, and “Edge with Delay”. When the typeis “State”, the oscilloscope triggers on the first edge when the qualified source is in the specified state(High or Low). When the type is “State with Delay”, a time limit condition is also available.
Qualified
SourceEdge
Trigger
SourceTrigger
Position

int.siglent.com

Qualified State = High
Edge = Rising103

SDS6000L User Manual

When the type is “Edge”, the oscilloscope triggers at the first edge after the specified edge (Rising or
Falling) of the qualified source; when the type is “Edge with Delay”, a time limit condition is available.

Qualified
SourceEdge
Trigger
SourceTrigger
Position

Qualified Edge = Rising
Edge = Rising

Click the Qualified Setting region to set the qualified source and threshold; Click the Edge
Setting region to set the edge trigger source, threshold, and slope.

13.5.12

Nth Edge Trigger

Nth edge trigger is similar to an edge trigger with conditions. The trigger is satisfied after meeting theuser-defined idle time and edge number conditions. As shown in the figure below, when the idle time
between the pulse trains is greater than the specified idle time, it is triggered on the third falling edgeof the pulse train.

Trigger source, slope (rising, falling), idle time, and edge number can be set in the trigger dialog box.
Holdoff, coupling, and noise reject can be set in the Nth edge trigger, see the sections "Holdoff","Trigger Coupling" and "Noise Reject" for details.

104

int.siglent.com

SDS6000L User Manual

13.5.13

Delay Trigger

A delay trigger is true when the edge of source B occurs after meeting the set conditions of source A
and a user-defined delay time. The setting of source A is similar to that of the pattern trigger and canbe used for logical "and" combination of multiple channels.

Click the SourceB Setting region to set the edge trigger source, threshold, and slope. Click theLimit Range and Upper / Lower Value region to set the delay time condition.

13.5.14

Setup/Hold Trigger

The clock and data sources need to be set in the setup/hold trigger setting. The set-up time starts when
the data signal crosses the trigger level and ends when the specified clock edge arrives. The holding
time starts when the specified clock edge arrives and ends when the data signal crosses the trigger
level again (as shown in the figure below). When the set-up time or hold time meets the preset time
limit conditions, the oscilloscope will trigger.

int.siglent.com

105

SDS6000L User Manual

Click the CLK Setting region to set the clock source, threshold, and slope; Click the Data Setting
region to set the data source, threshold, and state.

13.5.15

Serial Trigger

Refer to Chapter “Serial Trigger and Decode” for detailed information.

13.6

Trigger Source

The trigger sources supported by each trigger type are different. See the table below for details:
Trigger Type

106

C1~C8

EXT, EXT/5

AC LineD0~D15

Edge

√

√

√

√

Slope

√

×

×

×

Pulse

√

×

×

√

Video

√

×

×

×

Window√

×××

Interval

√

×

×

√

Dropout

√

×

×

√

Runt

√

×

×

×

Pattern

√

×

×

√

Serial√××√

Qualified

√

×

×

√

Nth Edge

√

×

×

√

Delay

√

×

×

√

Setup/Hold

√

×

×

√

int.siglent.com

SDS6000L User Manual

13.7

Holdoff

Holdoff is an additional condition for triggers and can be used to stabilize the triggering of complex
waveforms (such as a pulse series). It can be set to a time or number of events.

Holdoff by Time
Holdoff time is the amount of time that the oscilloscope waits before re-arming the trigger circuitry. The
oscilloscope will not trigger until the holdoff time expires.

Use the holdoff to trigger on repetitive waveforms that have multiple edges (or other events) between
waveform repetitions. You can also use holdoff to trigger on the first edge of a burst when you know
the minimum time between bursts.

For example, to achieve a stable trigger on the repetitive pulses shown in the figure below set theholdoff time (t) to 200ns<t<600ns.

Holdoff by Event
The Holdoff event sets the number of events that the oscilloscope counts before re-arming the trigger
circuitry. The oscilloscope will not trigger until the counter that tracks holdoff events reaches the set
value. In the following figure, the holdoff event is set to 3, and the signal is triggered on the fourth edge.

Parameter Start Holdoff On defines the initial position of holdoff.

int.siglent.com

107

SDS6000L User Manual

Acq Start -- The initial position of holdoff is the first-time point satisfying the trigger condition. In theexample above, each holdoff starts from the first rising edge of the pulse sequence.

Last Trig Time --The initial position of holdoff is the time of the last trigger. In the example above, thelast trigger position is at the second rising edge of the pulse sequence and the second holdoff starts
from that point.

13.8

Trigger Coupling

The coupling setting of a trigger is only valid when the trigger source is C1~C8, EXT, or EXT/5.


DC: All of the signal’s frequency components are coupled to the trigger circuit for high-frequency
bursts or where the use of AC coupling would shift the effective trigger level.



AC: The signal is capacitively coupled. DC levels are rejected. See the datasheet for details of the
cut-off frequency.



HFR: Signals are DC coupled to the trigger circuit and a low-pass filter network attenuates highfrequencies (used for triggering on low frequencies). See the datasheet for details of the cut-off
frequency.



LFR: The signal is coupled through a capacitive high-pass filter network, DC is rejected and low
frequencies are attenuated. For stable triggering on medium to high-frequency signals. See thedatasheet for details of the cut-off frequency.

108

int.siglent.com

SDS6000L User Manual

13.9

Noise Reject

Noise Reject adds additional hysteresis to the trigger circuitry. By increasing the trigger hysteresis, thenoise immunity becomes better but the trigger sensitivity degrades.

Noise Reject = Off

Noise Reject = On

int.siglent.com

109

SDS6000L User Manual

13.10 Zone Trigger
The SDS6000L includes a zone trigger to help isolate elusive glitches. There are two user-definedareas: Zone1 and Zone2. Users can set the property of each zone as “intersect” or “not intersect” asan additional condition to further isolate the interesting event quickly. “Intersect” only includes events
that occur within the zone. “Not-intersect” events include all events that occur outside of the zone.

Click Trigger > Zone to recall zone dialog box:
A.

Turn on or off zone trigger

B.

Select the source: C1~C8

C. Turn on or off zone1
D. Set the property of zone1: Intersect or Not
intersectE.

Set the coordinate of zone1. The range iswithin the waveform area

F.

Turn on or off zone2

G. Set the property of zone2: Intersect or Not
intersect
H. Set the coordinate of zone2. The range iswithin the waveform area

The zones can be created and moved by the mouse or by setting Zone > Zone Setting in the
dialog box. The color of the zone’s outline is consistent with the color of the specified source (Channel
1 = Yellow, etc...).

11 0

int.siglent.com

SDS6000L User Manual

Mouse
When the zone trigger is turned on, click-and-hold on any position within the waveform area and draw
a rectangular box, as shown in the following figure:

When the finger moves out of the screen, a menu pops up for selecting the zone and setting the zone
properties:

int.siglent.com

111

SDS6000L User ManualOnce a zone is created, it can be moved by dragging.

Select C1 as the source, turn on zone1,

Select C1 as the source, turn on zone1,and set the property as "Intersect"and set the property as "Not Intersect"

Dialog Box
Click Zone > Zone Setting to recall the dialog box.
A.

Set the left border of the zone

B.

Set the right border of the zone

C. Set the top border of the zone
D. Set the bottom border of the zone
E.

Return to the previous menu

Click the regions above to set the value by the mousewheel or the virtual keypad.

Note: If zone1 and zone2 are both turned on, the result of the "AND" operationin two zones becomes the qualified condition of triggering.

11 2

int.siglent.com

SDS6000L User Manual

Below is an example in which we want to capture a waveform of bus contention using the SDS6000L:

With a simple edge trigger, it is unlikely to trigger consistently on this anomaly. Thanks to the highwaveform update rate of the SDS6000L, we can confirm there is bus contention happening by enabling
the persistence display, as shown in the figure below:

int.siglent.com

11 3

SDS6000L User Manual

In this case zone trigger is a quick and simple way to capture the waveform. Enable the zone trigger,
and draw a box to intersect with the bus contention part, as shown in the figure below:

Now, we can accurately capture the exact bus contention waveform:

11 4

int.siglent.com

SDS6000L User Manual

14 Serial Trigger and Decode
14.1

Overview

The SDS6000L supports serial bus trigger and decode on the following serial bus protocols: I2C, SPI,UART, CAN, LIN, FlexRay, CAN FD, I2S, MIL-STD-1553B, SENT, and Manchester.

Click the trigger descriptor box, and then select the Type as Serial in the trigger dialog box toset the serial trigger:A.

Select the type as Serial

B.

Select the serial bus protocol

C. Set the signal, including the mapping
relation between channels and bus
signals, and the thresholds
D. Trigger setting
E.

Zone setting

int.siglent.com

11 5

SDS6000L User Manual

Perform Analysis > Decode to turn on the serial decode dialog box:
A.

Set the list of decoded results

B.

Select the bus to set, Bus1 and Bus2

C. Turn on/off the bus
D. Set the bus display format (Binary, Decimal,Hex, and ASCII)
E.

Select the serial bus protocol

F.

Click to set the signal, including the mapping
relation between channels and bus signals,
and the thresholds. This is similar to the signal
setting of a serial trigger.

G. Configure the bus protocol
H. Synchronize the settings between a serial
trigger and decode

Below are detailed descriptions of the trigger and decode steps for each protocol.

11 6



I2C Trigger and Serial Decode

SPI Trigger and Serial Decode

UART Trigger and Serial Decode

CAN Trigger and Serial Decode

LIN Trigger and Serial DecodeFlexRay Trigger and Serial Decode

CAN FD Trigger and Serial Decode

I2S Trigger and Serial Decode

MIL-STD-1553B Trigger and Serial Decode

SENT Trigger and Serial DecodeManchester Serial Decode

int.siglent.com

SDS6000L User Manual

14.2

I2C Trigger and Serial DecodeThis section covers triggering and decoding I2C signals. Please read the following for more details:"I2C Signal Settings", "I2C Trigger" and "I2C Serial Decode".

14.2.1I2C Signal Settings

Connect the serial data signal (SDA) and the serial clock signal (SCL) to the oscilloscope, set the
mapping relation between channels and signals, and then set the threshold level of each signal. The
signal settings of the decode and trigger are independent. If you want to synchronize the settings
between decode and trigger, please perform Copy Setting in the decode dialog box.

A.

Set the source of SCL. In the example above, SCL is connected to C4.

B.

Set the threshold level of SCL. It is 1.7V for the LVTTL signal in this example.

C. Set the source of SDA. In the example above, SDA is connected to C1.
D. Set the threshold level of the SDA channel.
E.

Return to the previous menu.

F.

Threshold level line. It only appears when adjusting the threshold level.

int.siglent.com

11 7

SDS6000L User Manual

Copy Setting
Click the Copy Setting in the decode dialog
box to synchronize the settings between thetrigger and decode.
A.

Copy the decode settings to trigger

B.

Copy the trigger settings to decode

C. Return to the previous menu.

Note: The synchronization is not automatic. If the settings in one placechange, a copy operation is necessary to re-synchronize.

14.2.2I2C Trigger

When the protocol is set to I2C, the following trigger conditions can be set: Start, Stop, Restart, No Ack,
EEPROM, or an R/W frame with a specific device address and data value.
Click Trigger Setting in the
I2C trigger dialog box to selectthe trigger condition:

Start -- The oscilloscope will be triggered when the SDA line transitions from high to low while the SCL
is high.
11 8

int.siglent.com

SDS6000L User Manual

Stop -- The oscilloscope will be triggered when the SDA line transitions from low to high while the SCL
is high.

Restart -- The oscilloscope will be triggered when another “Start” occurs before a “Stop".
No Ack -- The oscilloscope will be triggered when the SDA line is high during any SCL’s ACK bit.
EEPROM -- The trigger searches for EEPROM
control byte value 1010xxx on the SDA bus. And
there is a Read bit and an ACK bit behind
EEPROM. Set the data value and compare typeaccording to

Data1

and

Limit Range. If

EEPROM’s data is greater than (less than, equalto) Data1 , the oscilloscope will be triggeredat the edge of the ACK bit behind the Data byte.
The Data byte doesn’t need to follow the
EEPROM.

7 Address & Data -- the oscilloscope will be triggered on the read or write frame in 7-bit address mode.
Frame (Start: 7-bit address: R/W: Ack) -- Data1 and Data2 are set to “0xXX”. If all bits match, thentrigger on the Ack bit followed by the R/W bit.

int.siglent.com

11 9

SDS6000L User Manual

Frame (Start: 7-bit address: R/W: Ack: Data: Ack: Data2) -- If all bits match, then trigger on the Ack bit
followed by the Data2.

10 Address & Data -- If all bits match, then trigger on the Ack bit followed by the Data.
Frame (Start: Address 1st byte: R/W: Ack: Address 2nd byte: Ack: Data)120

int.siglent.com

SDS6000L User Manual

If you set the trigger condition to 7 address & data or 10
address & data:Address

can be selected in the hexadecimalrange of 0x00 to 0x7F (7-bit) or 0x3FF (10-bit). If theaddress is selected as "0xXX (7-bit address)" or"0xXXX (10-bit address)", the address is ignored. It
will always trigger on the Ack bit followed by theaddress.


Data1

and

Data2

can be selected in thehexadecimal range of 0x00 to 0xFF. If the data isselected as "0xXX", the data is ignored. It will
always trigger on the Ack bit followed by theaddress.


R/W Bit can be specified as Write, Read, or Don't
Care.

Data Length -- The data length range is 1 to 12 bytes. When the SDA data length is equal to the valueof Data Length and address length is the same as the set value, the oscilloscope will be triggered.
Click the Address Length to select "7-bits" or "10-bits" to match the address of the input signal.

Click the Data Length and roll the mouse wheel or use the virtual keypad to set the data length
to match the data length of the input signal.

int.siglent.com

121

SDS6000L User Manual

14.2.3I2C Serial Decode

The layout of the display when I2C decode is enabled is as follows:A.

The waveform display area shows the original waveforms of the bus signals

B.

The bus display shows the decoding result of the bus. At most two buses can be decoded at
the same time. Click Bus Operation in the dialog box to turn on or off the selected bus,
and click Bus Display to select the display type of decode result (Binary, Decimal, Hex, orASCII).

C. List display area. The decode result of multiple frames can be displayed in the list, in which
each row shows the time label and decode result of a frame. Click Result List to set theparameters of the list.
D. Decode dialog box

122

int.siglent.com

SDS6000L User ManualBus

The address value is displayed at the beginning of a frame. The write address is displayedin green and the read address is in yellow.



W/R bit is represented by (W) and (R), followed by the address value.



Theafter a data or address byte represents ACK (acknowledgment), andindicates no acknowledgment.
The data value is displayed in white.

A red point at the end of a segment indicates there is not enough space on the display toshow the complete content of a frame, and some content is hidden, such as.

The position of the bus in the vertical direction can be adjusted by dragging with the mouse.

List

TIME -- The horizontal offset of the current data frame head relative to the trigger position.



Address -- Address value. For example, "0x2AB" means that address = 2AB withacknowledgement.



R/W -- Read or write address.



DATA -- Data bytes.

int.siglent.com

123

SDS6000L User Manual

Configuration
There is only one item Include R/W Bit in the configuration of the I2C decode. When it is disabled,
the address is represented separately from the R/W bit, and when it is enabled, the R/W bit isrepresented together with the address.

For example, the address 0x4E: Write: Ack, is displayed as "0x4E (W)" when the R/W bit is not included
and is displayed as "0x9C" when the R/W bit is included.

124

int.siglent.com

SDS6000L User Manual

14.3

SPI Trigger and Serial DecodeThis section covers triggering and decoding SPI signals. Please read the following for more details:"SPI Signal Settings", "SPI Trigger" and "SPI Serial Decode".

14.3.1SPI Signal Settings

Connect the CLK, MOSI, MISO, and CS signals to the oscilloscope and set the mapping relation
between channels and signals. Then set the threshold level of each signal. The process of specifying
the source and threshold is similar to "I2C Signal Settings".

CLK
In addition to specifying the source and the threshold level, for the CLK signal, it is also necessary to
specify the Edge Select .
Rising–Data latched on the rising edge of the clock.



Falling–Data latched on the falling edge of the clock.

The user can select the edge according to the actual phase relationship between the clock and thedata of the SPI bus. Referring to the following figure, when the falling edge of the clock is aligned withthe data, the rising edge is selected to latch the data. When the rising edge of the clock is aligned withthe data, the falling edge is selected to latch the data.

int.siglent.com

125

SDS6000L User ManualCS
The CS signal should be set to correct CS Type , including CS, ~CS, and Clock Timeout.
CS – Active high. The CS signal needs a complete rising edge in the screen to be regarded as
active.

~CS – Active low. The ~CS signal needs a complete falling edge in the screen to be regarded as
active.



Clock Timeout – It is not necessary to specify the source and the threshold level for the CS signal.
The only parameter for the CS signal is the timeout Limit, which is the minimum time that the clock
signal must be held idle before the oscilloscope acquires valid data. This setting is suitable for thecase where the CS signal is not connected, or the number of oscilloscope channels is insufficient(such as two-channel oscilloscopes).

The method of copying settings is the same as I2C signal settings. See "I2C Signal Settings" for details.

Example:Connect the data, CLK, and ~CS signals of an SPI bus respectively to C1, C2, and C3. Data width =8-bit, Bit order = MSB, CS polarity = active low, and 12 data bytes are transmitted in one frame.

In the SPI trigger signal menu, set the source and threshold of CLK, MISO, and CS signals, then copy
the trigger settings to decoding. Adjust the timebase, so that there is a falling edge on the CS signal
shown on the screen:126

int.siglent.com

SDS6000L User Manual

When the CS type is set to Clock Timeout, turn on Cursor, measure the clock idle time between frames
as 150 us, and measure the interval between clock pulses as 1.28 us, then set the timeout to a value
between 1.28 us and 150 us. In this example it is set to 5 us:

If the data width is set to be greater than 8 bits (such as 16 bits), measure the clock idle time between
8-bit data packets as 11.28 us, and then set the timeout time to a value between 11.28 and 150 us. In
this example, it is set to 20 us:

int.siglent.com

127

SDS6000L User Manual

14.3.2SPI Trigger

The trigger condition for the SPI trigger is mainly about data.
Click Trigger Setting in the dialog box to set data:
A.

Trigger Type: MISO or MOSIB.

Set the trigger position type. When set to Any, it will
trigger on the specified data. When set to Value, itwill trigger on the data from the Start Bit

C. Set the start bit of the data value
D. Data Length: 4 ~ 96 bits when the start bit is 0
E.

Set to trigger on the specified data value. Click
Data Value twice, and enter the value by thevirtual keypad, or click All Same to set all bits to
0, 1, or ignored ("X")

F.

Set the bit order to MSB or LSB

G. Return to the previous menu

14.3.3SPI Serial Decode

The configuration of SPI decoding is similar to I2C.
In the Bus Config menu, Data Length (4-32 bit) and Bit Order (LSB or MSB) are configurable.

128

int.siglent.com

SDS6000L User Manual

UART Trigger and Serial Decode

14.4

This section covers triggering and decoding UART signals. Please read the following for more details:"UART Signal Settings", "UART Trigger" and "UART Serial Decode".

14.4.1UART Signal Settings

Connect the RX and TX signals to the oscilloscope, set the mapping relation between channels andsignals, and then set the threshold level of each signal. The process of specifying the source andthreshold is similar to "I2C Signal Settings".

In the BusConfig menu of trigger or decode, the following parameters are available:
A.

Click to select the baud rate: 600,1200, 2400,
4800, 9600, 19200, 38400, 57600, 115200 b/s
or Custom

B.

Data Length: 5~8 bit

C. Parity Check: None, Odd, Even, Mark, orSpace. If the data is 9 bits, the 9th bit is treatedas Mark parity or Space parity
D. Select the number of stop bits
E.

Set the idle levelF.

Set the bit order

G. Return to the previous menu

The method of copying settings is the same as I2C signal settings. See "I2C Signal Settings" for details.

int.siglent.com

129

SDS6000L User Manual

14.4.2

UART Trigger

Click Trigger Setting in the dialog box to set the trigger condition:A.

Source Type: RX or TX

B.

Trigger Condition: Start, Stop, Data or Error

C. When the "trigger condition" is Data, set thecompare type to =, >, <D. When the "trigger condition" is Data, set thedata value
E.

Return to the previous menu

Trigger ConditionStart -- The oscilloscope triggers when the start bit appears on the RX/TX.

Stop -- The oscilloscope triggers when the stop bits appear on the RX/TX. It always triggers on
the first stop bit.



Data -- Trigger on data✓

Click Compare Type to select "=", ">" or "<".

✓

Click Value to set the data value by the mouse wheel or virtual keypad. The range of data
values is 0x00 to 0xff.



Error -- The oscilloscope performs a parity check on the data according to the parity type set by
the user, and triggers if the check value is incorrect.

14.4.3

UART Serial Decode

The configuration of UART decoding is similar to that of I2C decoding.

130

int.siglent.com

SDS6000L User Manual

14.5

CAN Trigger and Serial DecodeThis section covers triggering and decoding CAN signals. Please read the following for more details:"CAN Signal Settings", "CAN Trigger" and "CAN Serial Decode".

14.5.1CAN Signal Settings

Connect the CAN_H and CAN_L signals to the oscilloscope, set the mapping relation between
channels and signals, and then set the threshold level of each signal. The process of specifying thesource and threshold is similar to "I2C Signal Settings".

In BusConfig menu of trigger and decode, baud rate can be set to: 5 kb/s, 10 kb/s, 20 kb/s, 50 kb/s,
100 kb/s, 125 kb/s, 250 kb/s, 500 kb/s, 800 kb/s, 1 Mb/s or Custom.

The method of copying settings is the same as I2C signal settings. See "I2C Signal Settings" for details.

14.5.2CAN Trigger

Click Trigger Setting in the CAN trigger dialog box to set the trigger condition:

Start -- The oscilloscope triggers at the beginning of the frame.



Remote -- The oscilloscope triggers on a remote frame with a specified ID. ID, ID Bits (11-bit or
29-bit), and Curr ID Byte (1st, 2nd, 3rd, or 4th byte) can be set. Curr ID Byte is used to specify
the byte to be adjusted when using the mouse wheel.

ID -- The oscilloscope triggers on the data frame that matches the specified ID. ID, ID Bits (11-bit
or 29-bit), and Curr ID Byte (1st, 2nd, 3rd, or 4th byte) can be set.

int.siglent.com

131

SDS6000L User ManualID + Data -- The oscilloscope triggers on the data frame that matches the specified ID anddata.ID, ID Bits (11-bit or 29-bit), Curr ID Byte (1st, 2nd, 3rd, or 4th byte), Data1, and Data2 canbe set.



Error -- The oscilloscope triggers on the error frame.

CAN Serial Decode

14.5.3The configuration of CAN decoding is similar to that of I2C decoding.

On the bus:

ID is displayed in green.

LEN (data length) is displayed in light yellow.



DATA are displayed in white.

CRC is displayed in blue.

The red point at the end of a segment indicates there is not enough space on the display toshow the complete content of a frame and some content is hidden.

In the list view:Time -- The horizontal offset of the current data frame head relative to the trigger position.

Type -- Type of the frame. The Data frame is represented by “D” and the remote frame isrepresented by “R”.

132

int.siglent.com

SDS6000L User ManualID -- ID of the frame, 11-bits or 29-bits ID are automatically detected.

Length -- Data length.



Data -- Data values.



CRC -- Cycle redundancy check.

Ack -- Acknowledge bit.

int.siglent.com

133

SDS6000L User Manual

14.6

LIN Trigger and Serial DecodeThis section covers triggering and decoding LIN signals. Please read the following for more details:“LIN Signal Settings”, “LIN Trigger” and “LIN Serial Decode”.

14.6.1

LIN Signal Settings

Connect the LIN signal to the oscilloscope, and then set the threshold level of the signal. The processof specifying the source and threshold is similar to “I2C Signal Settings”.
In BusConfig menu of trigger and decode, baud rate can be set to: 600 b/s,1200 b/s, 2400 b/s, 4800
b/s, 9600 b/s, 19200 b/s or Custom.

The method of copying settings is the same as I2C signal settings. See “I2C Signal Settings” for details.

14.6.2

LIN Trigger

Click Trigger Setting in the LIN trigger dialog box to set the trigger conditions:

Break -- The oscilloscope triggers at the beginning of the frame.

ID -- The oscilloscope triggers on the frame that matches the specified ID, which ranges from 0x00
to 0x3f.



ID & Data -- The oscilloscope triggers on the frame that matches the specified ID and data.ID,
Data1, and Data2 can be set.



134

Data Error -- The oscilloscope will trigger when a data error happens.

int.siglent.com

SDS6000L User Manual

LIN Serial Decode

14.6.3The configuration of LIN decoding is similar to that of I2C decoding.

On the bus:

ID is displayed in green

LEN (data length) and CHK are displayed in blue

DATA are displayed in white.

In the list view:Time -- The horizontal offset of the current data frame head relative to the trigger position.

ID -- ID of the frame.



Data length -- Data length.



ID Parity -- ID parity check.

Data -- Data values.



Checksum -- Data checksum.

int.siglent.com

135

SDS6000L User Manual

14.7

FlexRay Trigger and Serial DecodeThis section covers triggering and decoding FlexRay signals. Please read the following for more details:"FlexRay Signal Settings", "FlexRay Trigger" and "FlexRay Serial Decode".

14.7.1

FlexRay Signal Settings

Connect the FlexRay signal to the oscilloscope, and then set the threshold level of the signal. The
process of specifying the source and threshold is similar to "I2C Signal Settings".

In the BusConfig menu of the trigger and decode, the baud rate can be set to: 2.5 Mb/s, 5.0 Mb/s,
10.0 Mb/s, or Custom.

The method of copying settings is the same as I2C signal settings. See "I2C Signal Settings" for details.

14.7.2

FlexRay Trigger

Click Trigger Setting in the FlexRay trigger dialog box to set the trigger conditions:

TSS -- The oscilloscope triggers on the transmission start sequence.



Frame -- The oscilloscope triggers on the frame.
✓

Set Frame header indicators: Payload preamble indicator, null frame indicator, sync frame
indicator, startup frame indicator.

✓

Click ID to set the frame ID by the mouse wheel or virtual keypad. The range of ID is 0x000
to 0x7ff.

✓
136

Click Compare Type to select "=", ">" or "<". When setting the Compare Type as "=",
int.siglent.com

SDS6000L User Manual

the repetition factor is also required.
✓

Click Cycle to set the cycle count by the mouse wheel or virtual keypad. The range of data
values is 0 to 63.

Symbol -- The oscilloscope triggers on CAS/MTS (Conflict Avoidance Symbol/ Media access Test
Symbol) or WUS (Wake-up Symbol)



Errors -- The oscilloscope will trigger when a data error happens, including errors on FSS, BSS,
FES, Header CRC, and Frame CRC.

FlexRay Serial Decode

14.7.3The configuration of FlexRay decoding is similar to that of I2C decoding.

On the bus:The signatures (CAS/MTS, WUP) are displayed in yellow-green.



TSS transmission start sequence, displayed in yellow-green. The null frame indicator, theSync frame indicator, and the Startup frame indicator are displayed in the frame anddisplayed in pink.



The ID is displayed in the frame and is displayed in green.

PL (Valid Data Length) is displayed in frames, in words, and in light yellow.



HCRC (Head Check Code) is displayed in the frame and displayed in blue.



CYC (cycle) is displayed in the frame and displayed in light yellow.



D (data) is displayed in the frame and displayed in white.



FCRC (Data Check Code) is displayed in the frame and displayed in blue.

int.siglent.com

137

SDS6000L User Manual

In the list view:138Time -- The horizontal offset of the current data frame head relative to the trigger position.



FID -- Frame ID, the symbol occupies a single line of the list.

PL -- Valid Data Length

HCRC -- Head Check Code

CYC -- Cycle count

Data -- Data values



FCRC -- Data Check Codeint.siglent.com

SDS6000L User Manual

CAN FD Trigger and Serial Decode

14.8This section covers triggering and decoding CAN FD signals. Please read the following for more details:"CAN FD Signal Settings", "CAN FD Trigger" and "CAN FD Serial Decode".

14.8.1CAN FD Signal Settings

Connect the CAN FD signal to the oscilloscope, and then set the threshold level of the signal. The
process of specifying the source and threshold is similar to "I2C Signal Settings".

In BusConfig menu of trigger and decode, the nominal baud rate can be set to: 10 kb/s, 25 kb/s, 50
kb/s, 100 kb/s, 250 kb/s, 1 Mb/s or Custom. The data baud rate can be set to: 500 kb/s, 1 Mb/s, 2 Mb/s,
5 Mb/s, 8 Mb/s, 10 Mb/s or Custom.

The method of copying settings is the same as I2C signal settings. See "I2C Signal Settings" for details.

14.8.2CAN FD Trigger

Click Trigger Setting in the CAN FD trigger dialog box to set the trigger conditions:A.

Frame Type: Both, CAN, CAN FD

B.

When the "Frame Type" is CAN FD,
turn on or off ISO

C. When the "Frame Type" is CAN FD,
set the BRS (Bit Rate Switch) to Both,
Normal or FD
D. Trigger Condition: Start, Remote, ID,
ID + Data, Error
E.

Return to the previous menu

int.siglent.com

139

SDS6000L User Manual

Trigger ConditionStart -- The oscilloscope triggers at the beginning of the frame.



Remote -- The oscilloscope triggers on a remote frame with a specified ID. ID, ID Bits (11-bit or
29-bit), and Curr ID Byte (1st, 2nd, 3rd, or 4th byte) can be set. Curr ID Byte is used to specify thebyte to be adjusted when using the mouse wheel.

ID -- The oscilloscope triggers on the data frame that matches the specified ID. ID, ID Bits (11-bit
or 29-bit), and Curr ID Byte (1st, 2nd, 3rd, or 4th byte) can be set.



ID + Data -- The oscilloscope triggers on the data frame that matches the specified ID and data.ID,
ID Bits (11-bit or 29-bit), Curr ID Byte (1st, 2nd, 3rd, or 4th byte), Data1, and Data2 can be set.



Error -- The oscilloscope triggers on the error frame.
✓

Error Frame✓

Stuff Bit Error✓

CRC Mismatch Error: The oscilloscope triggers when the calculated CRC does not match thetransmitted CRC.

✓

Stuff Bit Cnt Err: Effective only when ISO is turned on, the oscilloscope triggers when the stuff
bit count is incorrect.

✓

Stuff Bit Cnt Par. Err: Effective only when ISO is turned on, the oscilloscope triggers when thepolarity of the stuff bit count is incorrect.

CAN FD Serial Decode

14.8.3The configuration of CAN FD decoding is similar to that of I2C decoding.
On the bus:

ID is displayed in green.

BRS (Bit Rate Switch) is displayed in light yellow.



ESI (Error State Indicator) is displayed in blue.

L (Data Length) is displayed in light yellow.



D (Data) is displayed in white.

CRC is displayed in blue.



Ack is displayed in pink.

The red point at the end of a segment indicates there is not enough space on the display toshow the complete content of a frame and some content is hidden.

140

int.siglent.com

SDS6000L User Manual

In the list view:Time -- The horizontal offset of the current data frame head relative to the trigger position.

Type -- Type of the frame. The Standard CAN frame is represented by "Std", CAN FD frame
is represented by "FD", the extended frame is represented by "Ext" and the remote frame isrepresented by "RTR".



ID -- Frame ID.

Length -- Data length.



Data -- Data bytes.

CRC -- Cycle redundancy check.

Ack -- Acknowledge bit.

int.siglent.com

141

SDS6000L User Manual

14.9

I2S Trigger and Serial DecodeThis section covers triggering and decoding I2S signals. Please read the following for more details:"I2S Signal Settings", "I2S Trigger" and "I2S Serial Decode".

14.9.1I2S Signal Settings

Connect the WS, BCLK, and Data signals to the oscilloscope, set the mapping relation between
channels and signals, and then set the threshold level of each signal. The process of specifying thesource and threshold is similar to "I2C Signal Settings"

BCLK
In addition to specifying the source and the threshold level, BCLK signals also require the specificationof the Edge Select .
Rising– Data latched on the rising edge of the clock.



Falling– Data latched on the falling edge of the clock.

WS
In addition to specifying the source and the threshold level, WS signals also require the specificationof the Left CH .


Low– Select the left channel when WS is low and the right channel when WS is high.



High– Select the right channel when WS is low and the left channel when WS is high.

The method of copying settings is the same as I2C signal settings. See "I2C Signal Settings" for details.

142

int.siglent.com

SDS6000L User Manual

I2S Trigger

14.9.2

Click Trigger Setting in the I2S trigger dialog box to set the trigger conditions :A.

Audio Variant: Audio-I2S, Audio-LJ,
Audio-RJ

B.

Trigger Condition: Data, Mute, Clip,
Glitch, Rising Edge, Falling Edge

C. Channel: Set the channel to trigger, Leftor Right
D. Set the bit order to MSB or LSB
E.

Channel Bits: Specify the receiver word
length. The range is from 1 to 32 bits

F.

Start Bit: The start bit of data, ranging
from 0 to 31

G. Data Bits: Specify the number of bits per
word. It should be less than the channel
bits
H. When the "Trigger Condition" is Data, set
the compare type to =, >, <I.

When the "Trigger Condition" is Data, set
the data value

J.

Return to the previous menu

Trigger Condition

Data -- Trigger on data.
✓

Click Compare Type to select "=", ">" or "<".

✓

Click Value to set the data value by the mouse wheel or virtual keypad. The range of data
value is related to the number of Data Bits .

Mute -- Trigger on the mute signal. Mute signal: Volume is less than the set value and duration
reaches the set value.

int.siglent.com

143

SDS6000L User Manual✓

Click MNF to set the mute threshold by the mouse wheel or virtual keypad. The range ofvalues is related to the number of Data Bits .

✓Click Duration to set the value by the mouse wheel or virtual keypad. The range is 1-64
frames.



Clip -- Trigger on the clipped signal. Clip signal: Volume is greater than the set value and duration
reaches the set value.
✓

Click Clip Level to set the clip threshold by the mouse wheel or virtual keypad. The rangeof values is related to the number of Data Bits .

✓Click Duration to set the value by the mouse wheel or virtual keypad. The range is 1-64
frames.



Glitch -- Trigger on glitches within the audio signal.
✓

Click Threshold to set the threshold by the mouse wheel or virtual keypad. The range ofvalues is related to the number of Data Bits .



Rising Edge -- Trigger on signals greater than the Threshold setting value.
✓

Click Threshold to set the threshold by the mouse wheel or virtual keypad. The range ofvalues is related to the number of Data Bits .



Falling Edge -- Trigger on signals less than the Threshold setting value.
✓

Click Threshold to set the threshold by the mouse wheel or virtual keypad. The range ofvalues is related to the number of Data Bits .

14.9.3I2S Serial Decode

The configuration of I2S decoding is similar to that of I2C decoding.
In the list view:Time -- The horizontal offset of the current data frame head relative to the trigger position.

Type -- Channel type, the left channel is represented by "Left CH" and the right channel isrepresented by "Right CH".

144



Data -- Data bytes



Error -- Error

int.siglent.com

SDS6000L User Manual

14.10 MIL-STD-1553B Trigger and Serial DecodeThis section covers triggering and decoding MIL-Standard 1553B signals. Please read the following
for more details: "MIL-STD-1553B Signal Settings" and "MIL-STD-1553B Serial Decode".

MIL-STD-1553B Signal Settings

14.10.1

Connect the MIL-STD-1553B signal to the oscilloscope, set the mapping relation between channels
and signals, and then set the threshold level of each signal. The process of specifying the source andthreshold is similar to "I2C Signal Settings"

MIL-STD-1553B Serial Decode

14.10.2

The configuration of MIL-STD-1553B decoding is similar to that of I2C decoding.
On the bus:
C/S the Command/Status word is displayed in green.



RTA the RT address is displayed in the frame and displayed in light yellow.

C/S Data is displayed in the frame and is displayed in white.

The data word is displayed in yellow and the data is displayed in the frame and displayed inwhite.

The Check Code is displayed in the frame and displayed in blue.

In the decoding list view:Time— The horizontal offset of the current data frame head relative to the trigger position.



RTA— The RT address

int.siglent.com

145

SDS6000L User Manual

Type— Type of the word

Data— Data values

Error— Error type14.11 SENT Trigger and Serial DecodeThis section covers triggering and decoding SENT signals. Please read the following for more details:"SENT Signal Settings", "SENT Trigger" and "SENT Serial Decode".

14.11.1

SENT Signal Settings

Connect the SENT signal to the oscilloscope, set the mapping relation between channels and signals,
and then set the threshold level of each signal. The process of specifying the source and threshold issimilar to "I2C Signal Settings"
In the BusConfig menu of trigger or decode, the following parameters are available:
A.

Click to specify the nominal clock period (tick) timeB.

Set the percent tolerance to determine whether the sync
pulse is valid for decoding the data

C. Set the number of nibbles in a Fast Channel Message
D. Set the CRC format that will be used in calculating thecorrectness of the CRCs. The New CRC selection uses the2010 CRC format. If “NEW” is not selected, the CRC will
use the 2008 format. Enhanced Serial Message CRCs arealways calculated using the 2010 format, but for the Fast
Channel Messages, and Short Serial Message CRCs, thechosen setting is used
E.

Specify whether there is a pause pulse between Fast
Channel Messages

F.

146

Return to the previous menu

int.siglent.com

SDS6000L User Manual

14.11.2

SENT Trigger

When the protocol is set to SENT, the following trigger conditions can be set: Start, Slow Channel, Fast
Channel, and Error.
Click Trigger Setting in the
SENT trigger dialog box toselect the trigger condition:

Start -- The oscilloscope will be triggered at the start of the message (after 56 synchronization ticks).
You can select the type of message: Fast Channel Message, Slow Channel Message, or any.

Fast Channel -- The oscilloscope will be triggered on a Fast Channel Message when the Status &Communication nibble and the data nibbles match the specified values.
State can be selected in the hexadecimal range of 0x0
to 0xF. If the value is selected as "0xX", the state isignored.

Compare Type can be specified as Equal, Not Equal,
Less than, Less or Equal, Greater than, Greater or Equal,in Range of, or out of Range.



At Pos determines the start position of the trigger data.
If selected as Value, you need to set Nibble Pos in thedecimal range of 0 to 5. If selected as Don’t care, theoscilloscope will be triggered at the first data whichmatches the specified condition.



Nibble Len can be selected in the decimal range of 1 to
6. It is associated with the Nibble Pos .



Data can be selected in hex. The range is associatedwith the Nibble Pos . If the data is selected as "0xXX",
the data is ignored.

int.siglent.com

147

SDS6000L User Manual

Slow Channel -- the oscilloscope will be triggered on a Slow Channel Message.
Frame (short serial message) -- The 16-bit message consists of a 4-bit Message ID nibble, 2 nibbles(1 byte) of data, and a CRC checksum nibble. If ID and data match, then trigger on the end of CRC bit.

The serial message frame contains 21 bits of payload data. Two different configurations can be chosen
and determined by the configuration bit (serial data bit #3, serial communication nibble No. 8):

Frame (enhanced serial message with 4bits ID) --16-bit data and 4-bit message ID, the configuration
bit is 1.

Frame (enhanced serial message with 8bits ID) —12-bit data and 8-bit message ID, the configuration
bit is 0.

148

int.siglent.com

SDS6000L User Manual

If you set the trigger condition to Slow Channel:

ID can be selected in the hexadecimal range of 0x0 to
0xF (short serial / enhanced serial with 4 bits ID) or 0x00 to
0xFF (enhanced serial with 8 bits ID). If it is selected as"0xXX", the ID is ignored.

Compare Typecan be specified as Equal, Not Equal,

Less than, Less or Equal, Greater than, Greater or Equal,in Range of, or out of Range.
Data can be selected in the hexadecimal range of 0x00
to 0xFF (short serial) or 0x0000 to 0xFFFF (enhanced serial
with 4 bits ID) or 0x000 to 0Xfff (enhanced serial with 8 bits
ID). If it is selected as "0xXX", the data is ignored.

Error -- The oscilloscope triggers on the error frame. Errors include Successive Sync Pulses Error,
Pulse Period Error, Fast Channel CRC Error, Slow Channel CRC Error, All CRC Errors.
Successive Sync Pulses Error: Triggers on a sync pulse whose width varies from the previous
sync pulse's width by greater than 1/64 (1.5625%, as defined in the SENT specification).



Pulse Period Error: Triggers if a nibble is either too wide or too narrow (for example, data nibble <12 (11.5) or > 27 (27.5) ticks wide). Sync, S&C, data, or checksum pulse periods are checked.



Fast Channel CRC Error: Triggers on any Fast Channel Message CRC error.



Slow Channel CRC Error: Triggers on any Slow Channel Message CRC error.



All CRC Errors: Triggers on any CRC error, Fast or Slow.

int.siglent.com

149

SDS6000L User Manual

SENT Serial Decode

14.11.3

The configuration of SENT decoding is similar to that of I2C decoding.
On the bus:
For Fast Channel:


SYNC is displayed in pink

STATE is displayed in green

DATA is displayed in white

CRC and Pause pulse are displayed in blue

For Slow Channel:

150

ID is displayed in green

DATA are displayed in white

CRC is displayed in blue

int.siglent.com

SDS6000L User Manual

In the list view:Time — The horizontal offset of the current data frame head relative to the trigger position.

Sync — Sync pulse width (only fast channel)

State — Status & Communication nibble (only fast channel)ID — ID of the frame (only slow channel).



Data — Data values

CRC — Cycle redundancy checkPause — Pause ticks

Error — Error type14.12 Manchester Serial DecodeThis section covers decoding Manchester signals. Please read the following for more details:"Manchester Signal Settings" and "Manchester Serial Decode".

int.siglent.com

151

SDS6000L User Manual

14.12.1

Manchester Signal Settings

Connect the Manchester signal to the oscilloscope, set the mapping relation between channels andsignals, and then set the threshold level of the signal. The process of specifying the source andthreshold is similar to "I2C Signal Settings"
In the BusConfig menu of decode, the following parameters are available:
A.

Click to specify the baud rate, the range is 500
b/s to 5 Mb/s

B.

Set the Manchester signal's logic type. R:1
indicates that the rising edge is used to encode
a bit value of logic 1, and F:1 indicates that thefalling edge is used to encode a bit value oflogic 1

C. Set the idle level
D. Set the starting edge of the Manchester signal.
The range is 1~32
E.

Set the minimum idle time/inter-frame gap timeof the Manchester bus in terms of the bit width.

F.

Set the display format to Byte or Bit

G. Set the bit order to MSB or LSB
H. Set the sync field size, from 0 to 32
I.

Set the header size, from 0 to 32

J.

Set the number of words in the data field, from
1 to 255

K.

Set the data word size, from 2 to 8

L.

Set the trailer size, from 0 to 32

M. Return to the previous menu

152

int.siglent.com

SDS6000L User Manual

Manchester Serial Decode

14.12.2

The configuration of Manchester decoding is similar to that of I2C decoding.

On the bus:

SYNC is displayed in pinkThe header is displayed in green

DATA is displayed in whiteThe trailer is displayed in blue

In the list view:Time -- The horizontal offset of the current data frame head relative to the trigger position.



Data -- Data words



Error -- Error typeint.siglent.com

153

SDS6000L User Manual

15 Cursors
15.1

Overview

Cursors are important tools when measuring signals. Rapid measurements can be performed using
cursors in both horizontal and vertical directions. The cursor types include X1, X2, X1-X2, Y1, Y2, andY1-Y2, used to indicate X-axis values (time or frequency) and Y-axis values (amplitude) on a selected
waveform (C1~C8/F1~F4/M1~M4/REF/Histogram).

Perform Cursors > Menu to open the cursors dialog box:
A.

Turn the cursor function on or off

B.

Cursor Mode. The vertical cursors will automaticallytrack the waveform in Trace mode.

C. Specify the cursor
D. Set the position of the specified cursor
E.

Select the cursors type (horizontal, vertical, horizontal +vertical), this option is supported only in "Manual" mode.

F.

Select the source

G. Display style of cursors text
H. X cursor reference (Delay or Position)
Delay measures the cursor position relative to the
Horizontal Delay/Offset position
Position measures the cursor position relative to thehorizontal center of the display
I.

Y cursor reference (Offset or Position), this option is
supported only in "Manual" mode.
Offset measures the cursor position relative to the
Vertical/Offset position
Position measures the cursor position relative to thevertical center of the display

154

int.siglent.com

SDS6000L User Manual

Cursors Mode

Manual -- Manually sets the cursors’ position. Cursors type (horizontal, vertical, horizontal +vertical) is available in this mode.

Track -- The cursor type is automatically set to "horizontal + vertical". In this mode, only horizontal
cursors are adjustable, while the vertical cursors automatically attach to the cross-point of the
cursor and the source waveform.

Measure – Automatically indicates the measured item using cursors

Manual Mode

Track Mode

int.siglent.com

155

SDS6000L User Manual

Measure Mode

Cursors TypeX (horizontal) -- Vertical dotted lines that measure horizontal time (when the source is an FFT waveform,
X cursors measure frequency).

X cursors (time)

X cursors (frequency)

X1 -- The left (default) vertical dotted line. It can be manually moved to any horizontal positionon the screen.

X2 -- The right (default) vertical dotted line. It can be manually moved to any horizontal
position on the screen.



X1- X2 -- The difference between X1 and X2. After this option is selected, changing the value
will move both X1 and X2 simultaneously.

156

int.siglent.com

SDS6000L User ManualY（vertical）-- Horizontal dotted lines that measure vertical voltage or current (depending on the unitof the selected channel). When the cursor's source is the math function, the units will match the mathfunction.



Y1 -- The upper (default) horizontal dotted line. It can be manually moved to any vertical
position on the screen.

Y2 -- The lower (default) horizontal dotted line. It can be manually moved to any vertical
position on the screen.

Y1- Y2 -- The difference between Y1 and Y2. After this option is selected. Change the value
to move both Y1 and Y2 simultaneously.

X+Y (horizontal + vertical) -- Both the X cursors and Y cursors are enabled.

int.siglent.com

157

SDS6000L User Manual

Display Mode

Display Mode M1Display Mode M2

M1 -- The position information of each cursor is attached to the cursor, and the difference
information is between the two cursors with arrows connected to the cursors. This mode is moreintuitive.



M2 -- The position information of each cursor and the difference between the cursors are displayed
in a region on the screen. The region can be moved by dragging to avoid covering the waveform.
This mode is relatively concise.

Cursors Reference
X cursors reference:

Fixed Delay– When the timebase is changed, the value of X cursors remains fixed.



Fixed Position– When the timebase is changed, the X cursors remain fixed to the grid positionon the display.

Y cursors reference:

Fixed Offset– When the vertical scale is changed, the value of

Y- cursors remain fixed.


Fixed Position– When the vertical scale is changed, the Y-cursors remain fixed to the grid
position on the display.

158

int.siglent.com

SDS6000L User Manual

Take the X cursors reference as an example to demonstrate the scaling effect of different settings:

Timebase=50ns/div，X1=-50ns=-1div，X2=100ns=2div

Fixed position, timebase is changed to 100 ns/div, and the grid number of X cursors(-1div, 2div) remains fixed. The value of X1 and X2 is changed to -100 ns, 200 ns.

int.siglent.com

159

SDS6000L User ManualFixed delay, timebase is changed to 100 ns/div, and the value of X cursors (-50 ns, 100 ns)
remains fixed. The grid number of X cursors is changed to -0.5div, 1div.

15.2

Select and Move Cursors

The cursors can be selected and moved directly by the mouse, in addition, they can be selected in the
cursor’s value dialog box.

160

int.siglent.com

SDS6000L User Manual

Mouse
Directly click the cursor and drag it, as shown below:

Click the display area of △X (or △Y) in M1 mode and drag it to move the two cursors simultaneously,
as shown in the figure below. This is equivalent to the operation on the cursor type X1-X2 or Y1-Y2.

Dialog Box
Click the cursor value area of the dialog box, and then roll themouse wheel to adjust the position.

int.siglent.com

161

SDS6000L User Manual

16 Measurement
16.1

Overview

The SDS6000L features a strong automatic measurement list. These parameters can be automaticallymeasured without cursors and include common measurements such as rise time, fall time, peak-peak,
and period. The SDS6000L can also measure multiple channels at the same time, showing up to 6
parameter measurements with statistics while in the M1 display mode and up to 12 parameters in theM2 mode. If you wish to view more parameters on a specified channel, then the "Simple" mode can be
employed. For measuring waveform of interest in a time gate, the "Gate" function is suggested.
Some parameter measurements (such as the mean) may be a value generated by all the data in aframe. Some parameter measurements (such as period) accumulate all measurements in a frame, but
the displayed value is always the first value. If you want to know the distribution of multiple parameters
in one frame, you should use the statistics function.

A.

The waveform display area automatically compresses when the other windows are displayed

B.

Measurement parameters and statistics display area. If select the mode as “Simple”, the "Simple"
parameter area is displayed

C. Statistics histogram display area
D. Measure dialog box

162

int.siglent.com

SDS6000L User Manual

Perform Measure > Menu to open the dialog box.
A.

Enable/disable measureB.

Set the mode of measure: Simple or Advanced.
"Simple" shows the specified basic measurement
parameters of the selected channel. In "Advanced"
mode, the measurement parameters can be added
one by one as needed

C. Measure configuration: gate, amplitude strategy,
threshold, and display mode
D. Clear all the selected measurements
E.

Select measurement parameters

F.

Tools including Trend, Track, and Measure Cursor

G. Turn on or off statistics
H. Statistics settings: count limit, reset statistics, andhistogram

16.2

Set Parameters

Click Type in the measure dialog box, or click + in the measurement parameters and statistics
display area to open the parameter selection window:

int.siglent.com

163

SDS6000L User ManualA.

Set the source of the current setting.

B.

Measurement parameter classification tabs, including Favorite, Vertical, Horizontal,
Miscellaneous, and CH Delay. Click a tab and thearea will display the corresponding

parameters
C. Parameters. Click the parameter to be measured to activate it, and click it again to close theparameter.
D. Background highlighted parameters represent it is activated. In the figure above, "Pk-Pk" and"ROV" are activated.
E.

Description of the last selected parameter.

The correct steps to add a measurement parameter are to select the source in thethen select the parameter in thearea andarea. For example, to add Pk-Pk measurements for C1 andPeriod measurements for C2, follow the steps below:
Source > C1 > Vertical > Pk-Pk
Source > C2 > Horizontal > Period

For the channel delay (CH Delay) measurement, because the number of sources involved is greaterthan 1, the steps to specify the source are different:

In the parameter selection area, the channel corresponding to Source A is specified first, and thenthe channel corresponding to

Source B . Finally, the measurement parameter is selected. For

example, to activate the skew between C1 and C2, you can follow the following steps:
Source A > C1 > Source B > C2 > Skew

164

int.siglent.com

SDS6000L User ManualOnce a parameter is selected, it will appear in the parameter and statistical display area below the grid:

Click + in the blank area to add a parameter.
Click – in the upper right corner of each parameter to close the parameter.
Click × in the upper right corner of the area to close the measurement.
Click Clear in the dialog box to close all parameters.

The “Favorite” tab is used to store frequently used items. You can customize this tab. Up to 20 items
can be stored in it. Long-click an item to add or remove it from the “Favorite” tab. For example, to add
Pk-Pk to the “Favorite” tab:
Advanced > Type > Vertical > Pk-Pk > Add to favorite

To remove Period from the “Favorite” tab:
Advanced > Type > Favorite > Period > Delete from favorite

int.siglent.com

165

SDS6000L User Manual

16.3

Type of Measurement

16.3.1

Vertical Measurement

Vertical measurement includes 19 parameters:

Max: Highest value in the input waveform

Min: Lowest value in the input waveformPk-Pk: Difference between maximum and minimum data values

Top: Value of most probable higher state in a bimodal waveform

Base: Value of most probable lower state in a bimodal waveform

Amplitude: Difference between top and base in a bimodal waveform. If not bimodal, thedifference between max and min166

Mean: Average of data values



Cycle Mean: Average of data values in the first cycle

Stdev: Standard deviation of the data

Cycle Stdev: Standard deviation of the data in the first cycle
int.siglent.com

SDS6000L User Manual

RMS: Root mean square of the data

Cycle RMS: Root mean square of the data in the first cycleMedian: Value at which 50% of the measurements are above and 50% are below



Cycle Median: Median of the first cycleOvershoot (FOV): Overshoot following a falling edge; 100%* (base-min)/amplitudeOvershoot (ROV): Overshoot following a rising edge; 100%*(max-top)/amplitude

Preshoot (FPRE): Overshoot before a falling edge.
Equal to 100 %*( max-top)/amplitude.

Preshoot (RPRE): Overshoot before a rising edge.
Equal to 100 %*( base-min)/amplitude.

int.siglent.com

167

SDS6000L User Manual

L@T: Level measured at trigger position

Horizontal Measurement

16.3.2

Horizontal measurement includes 17 parameters:

Period: Time between the middle threshold points of two consecutive like-polarity edges.



Freq: Reciprocal of the period

Time@max: First time of maximum valueTime@min: First time of minimum value+Width: Time difference between the middle threshold of a rising edge to the middle
threshold of the next falling edge of the pulse

168

int.siglent.com

SDS6000L User Manual-Width: Time difference between the middle threshold of a falling edge to the middle
threshold of the next rising edge of the pulse+Duty: Positive Duty Cycle. The ratio of positive width to period-Duty: Negative Duty Cycle. The ratio of negative width to period+BWidth: Time from the first rising edge to the last falling edge at the middle threshold-BWidth: Time from the first falling edge to the last rising edge at the middle threshold

Delay: Time from the trigger to the first transition at the middle thresholdT@M: Time from the trigger to each rising edge at the middle threshold

Rise Time: Duration of rising edge from lower threshold to upper threshold

Fall Time: Duration of falling edge from upper threshold to lower threshold

10-90%Rise: Duration of rising edge from 10-90%90-10%Fall: Duration of falling edge from 90-10%

CCJ: The difference between two continuous periods

Miscellaneous Measurements

16.3.3The Miscellaneous measurements tab includes 16 parameters:+Area@DC: Area of the waveform above zero-Area@DC: Area of the waveform below zeroArea@DC: Area of the waveform

AbsArea@DC: Absolute area of the waveform+Area@AC: Area of the waveform above average-Area@AC: Area of the waveform below average

int.siglent.com

169

SDS6000L User ManualArea@AC: Area of the waveform above the average minus the area of the waveform
below average

AbsArea@AC: Area of the waveform above the average plus the area of the waveform
below the averageCycles: Number of cycles in a periodic waveform

Rising Edges: Number of rising edges in a waveform

Falling Edges: Number of falling edges in a waveform

Edges: Number of edges in a waveform

Ppulses: Number of positive pulses in a waveform



Npulses: Number of negative pulses in a waveform

PSlope: The slope of the rising edgeNSlope: The slope of the falling edge

Delay Measurement

16.3.4

Delay measurement measures the time difference between two channels. It includes 14 delay
parameters:Phase: Phase difference between two edges



Skew: Time of source A edge minus time of nearest source B edge



FRFR: The time between the first rising edge of source A and the following first risingedge of source B at the middle threshold



FRFF: The time between the first rising edge of source A and the following first fallingedge of source B at the middle threshold170

int.siglent.com

SDS6000L User Manual

FFFR: The time between the first falling edge of source A and the following first risingedge of source B at the middle threshold



FFFF: The time between the first falling edge of source A and the following first fallingedge of source B at the middle threshold

FRLR: The time between the first rising edge of source A and the last rising edge ofsource B at the middle threshold

FRLF: The time between the first rising edge of source A and the last falling edge ofsource B at the middle threshold



FFLR: The time between the first falling edge of source A and the last rising edge ofsource B at the middle threshold

FFLF: The time between the first falling edge of source A and the last falling edge ofsource B at the middle threshold

16.4



Tsu@R: Data setup time before the clock rising edge



Tsu@F: Data setup time before the clock falling edgeTh@R: Data hold time after the clock rising edge

Th@F: Data hold time after the clock falling edge

Trend

After adding a measurement parameter, a Trend can be used to observe the long-term change of theselected measurement value over time.

int.siglent.com

171

SDS6000L User Manual

A.

Measurement parameter display area

B.

Trend plot display area

C. Extend the range of time. Click it to expand the time range.
D. Current valueE.

Maximum value

F.

Minimum value

G. Average value
H. Reset statistics

Click Reset Statistics in the measure dialog box, or click the symbolin the statistics display

area to clear and restart statistics.

16.5

Track

The measure values VS. time plot of a horizontal parameter (e.g. frequency, rise time) in one frame
can be observed when the Track is enabled.

The upper limit of the statistic number in a frame is set in Statistics > AIM Limit , which means
the values exceeding the limit will not be shown in the track plot. The upper limit is 1,000 by default
and can be set to up to 65,000.

172

int.siglent.com

SDS6000L User Manual

16.6

Display Mode

In advanced measurement mode, two display modes are supported: M1 and M2.
In M1 mode, up to 6 parameter measurements are displayed at a time. When statistics are enabled,
they are listed under the measurement items. Click a column to add or replace a measurement.

In M2 mode, up to 12 parameter measurements are displayed at a time. When statistics are enabled,
they are distributed on the right side of the measurement item. Click a row to add or replace ameasurement.

int.siglent.com

173

SDS6000L User Manual

Note: In M2 mode histogram display is not supported.

16.7

Measurement Statistics

Measurement statistics are based on the total number of captured waveforms. In Roll mode,measurement statistics increase over time. Click Statistics Setting in the measure dialog box torecall the Statistics Config dialog box:
A.

Set the maximum number of samples for thestatistics function. The setting range is 0 ~ 1024, or
unlimited. If there is no limit, the number of statistics
will be accumulated. If there is a limit, when the
maximum number of statistics N is reached, only thelatest N measurements will be counted. When the
count is limited, only the first period measurement
value of each frame is counted

B.

Set the upper limit of the statistic number in a frame.
Valid only when the count limit is set to "Unlimited".
The setting range is 1 ~ 65,000

C. Clear and restart the statistics.
D. Turn the histogram on or off
E.

Return to the previous menu

Enable the Statistics function to observe the distribution of the measured values of every selectedparameter.

174

int.siglent.com

SDS6000L User Manual

Value – The current measurement

Mean – Average of all historical measurements

Min – The minimum of all historical measurements

Max – The maximum of all historical measurements



Stdev – Standard deviation of all historical measurements

Count – The number of historical measurements

Click Reset Statistics in the statistics setting dialog box, or click the symbolin the statistics

display area to clear and restart statistics.

When the oscilloscope detects the trace under test is clipped, an extra overflow indicator will appear
after the measurement value:
Waveform clipped at the top
Waveform clipped at the bottom
Waveform clipped at both top and bottom

16.8

Statistics Histogram

After enabling statistics on a selected measurement, you can also activate a statistical histogram. The
histogram appears at the bottom of the statistics area. This enables users to quickly view the probability
distribution of the measured parameters. The color of the histogram is consistent with the measurement
source (Channel 1 is yellow, etc.).

int.siglent.com

175

SDS6000L User Manual

Click the histogram area of a parameter to enlarge it for details. You can move the large histogram
window position around the display by dragging. Click the histogram of another parameter to switch tothe corresponding enlarged histogram.

A.

Parameter

B.

Histogram display area. The X-axis represents measured values and Y-axis represents theprobability.

C. Count of statistics
D. Current value
E.

The bin that includes the maximum value and the probability of a value falling into it

F.

Close the enlarged histogram

16.9

Simple Measurements

Enabling Simple Measurement displays all selected measurement parameters of the specified channel
at the same time. The font color of the measurement parameters is consistent with the color of thespecified source. Yellow for Channel 1, Purple for Channel 2, etc.

176

int.siglent.com

SDS6000L User Manual

16.10 Gate
Sometimes the user may want to measure parameters for a certain specified time range of the signal
and ignore signal parts that lie outside of that range. In this case, the Gate function can be helpful.

Turn on the Gate, then two horizontal cursors A and B willappear in the grid area. The gate setting dialog box will be
displayed on the right.
Gate cursors A and B are used to define the range of time forparameter measurements. The oscilloscope only measures
the parameters of the data between A and B, ignoring thedata outside the range.
The setting of gate cursors is similar to that of ordinary
cursors. See "Select and Move Cursors" for details.

The figure below shows a scenario in which the gate function is used to measure the peak-peakparameter of the trough of an amplitude-modulated waveform:

int.siglent.com

177

SDS6000L User Manual

16.11 Amplitude Strategy
According to different types of input signals, users can choose the corresponding amplitude calculation
strategy which can measure top and bottom values with more accuracy.
Perform Config > Amplitude Strategy in the measure dialog box to recall the amplitude strategy
dialog box:
A.

Set the amplitude calculation strategy. When set
to Auto, the amplitude calculation strategy will beselected automatically according to the input
signal to ensure the accuracy of the measured
value

B.

Set the top value calculation strategy. When set
to Histogram, the value at the upper half of the
waveform will be counted, and the value with the
maximum probability will be identified as the top
value; when set to max, the maximum value ofthe waveform will be identified as the top value

C. Set the base value calculation strategy. When set to Histogram, the value at the lower
half of the waveform will be counted, and the value with the maximum probability will beidentified as the base value; when set to min, the minimum value of the waveform will beidentified as the base value
D. Return to the previous menu

16.12 Threshold
Measurement thresholds can be defined by the user. This is more flexible than fixed thresholds. For
example, for pulse-width measurement, the threshold can be specified rather than fixed at 50%. For
rise time, the lower / upper thresholds can be specified rather than fixed at 10% / 90%.

Changing the default threshold may change the measurement results of relevant measurement items,
such as Period, frequency, +width, -width, +duty, -duty, +BWidth, -BWidth, delay, T@M, rise time, falltime, CCJ, cycles, rising edges, falling edges, edges, Ppulses, Npulses, and delay measurement.

Perform Config > Threshold in the measure dialog box to recall the threshold dialog box:
178

int.siglent.com

SDS6000L User ManualA.

Set measurement threshold source

B.

Set the type of threshold

C. Set the upper value
D. Set the middle valueE.

Set the lower value

F.

Return to the previous menu

Threshold Type
Percent: Set according to the percentage of the waveform. The setting range of lower value and upper
value is 1% ~ 99%, and the low value shall not be greater than the middle value and high value
Absolute: Set according to the vertical range. The absolute threshold value depends on the verticalscale, offset and probe attenuation. These values should be set before setting absolute thresholds.
Lower and upper values are limited to the range of the screen. If any absolute threshold is greater orless than the minimum or maximum waveform value, the measurement may be invalid.

16.13 Hardware Frequency Counter
The hardware frequency counter is a stand-alone, hardware-based frequency measurement function.
Differing from the “Frequency” parameter (see section “Horizontal Measurement”) which is measured
by software, the hardware frequency counter directly uses the trigger signal as the input of a hardware
counter, which yields 7-digit frequency measurement accuracy. The accuracy of the “Frequency”
parameter measured by software is relative to the sample rate. The lower the sample rate, the worse
the accuracy. In situations where the sample rate is less than 2 times the input frequency, the results
will be incorrect. The hardware frequency counter does not have this limitation. Its result is always
shown at the top-right of the display as shown below:

int.siglent.com

179

SDS6000L User Manual

17 Math17.1

Overview

The SDS6000L supports 4 math traces and multiple operators. Arithmetic operators: Addition (+),
subtraction (-), multiplication (x), division (/), average, ERES, identity, negation, maxhold, minhold.
Algebra operators: differential (d/dt), integral (∫dt), square root (√), absolute (|y|), sign, exp, ln,interpolate; and FFT, Filter, as well as a formula editor. The math traces are labeled with icons “F1 ~F4” and can be measured by the cursors or as the source of auto measurement.

Click + in the channel descriptor box region and select F1 , F2 , F3 , or F4 , and thenthe math dialog box pops up.
A.

Select the math trace, F1-F4

B.

Turn on or off the math operation

C. Select the function. Click the area to recall thefunction setting page, and select the source andoperator
D. Set the vertical scale of the math operationE.

Set the label text of the math traceF.

Set the vertical position of the math operation,referring to "Vertical Setup"

G. Enable/Disable Invert, similar to the process usedin "Vertical Setup".

Units for Math Waveform
Different operations have different dimensions. Therefore, the specific units displayed in Scaledepend on the operation:180

int.siglent.com

SDS6000L User ManualMath Operation

Unit

Addition (+) or

V, A, or U*

Subtraction (-)*(used when the units of two sources are not consistent)

Multiplication (x)V∧2, A∧2, or W

Division (/)

None, Ω (Resistance unit Ohms), S (conductance unit Siemens)

FFTdBVrms, Vrms, dBArms, Arms, dBm

d/dt

V/s (Volt/second) or A/s (A/second)∫dt

VS (Volt*second) or AS (A*second)√V ∧ 0.5 or A ∧ 0.5

|y|

V, A

Sign

V, A

Exp or Exp10

V, A

Ln or lg

V, A

intrp

V, A

17.2

Arithmetic

The SDS6000L supports Addition, Subtraction, Multiplication, Division, Average, ERES, Identity,
Negation, Max-hold, and Min-hold.

17.2.1

Addition / Subtraction / Multiplication / Division

The SDS6000L can perform arithmetic operations including addition, subtraction, multiplication, or
division on any two analog input channels, and the values of Source A and Source B are computed
point-by-point.

int.siglent.com

181

SDS6000L User Manual

The following figure shows an example of F1 = C1 + C2:

17.2.2

Identity / Negation

The values of Source A are computed point-by-point when an identity or negation operation is chosen
in any analog channel. The following figure shows an example of F1 = -C1:

182

int.siglent.com

SDS6000L User Manual

Average / ERES

17.2.3

Average / ERES can be set in acquisition mode and arithmetic function as well, but with different
computing methods. Average and ERES in acquisition mode are computed by hardware with a higher
speed than if they were computed by software in the arithmetic mode. Average and ERES in acquisitionmode can only be based on data from analog channels, while the Average / ERES arithmetic function
can use analog channels, zoom traces, math, and memory traces as the source.

Max-hold / Min-hold17.2.4The value of Max-hold and Min-hold are based on statistical calculations from multiple frames. The
following figure shows the Max-hold result.

17.3

AlgebraThe SDS6000L can perform algebraic operations including differential (d/dt), integral (∫dt), square root(√), absolute(|x|), sign, exp (𝑒 𝑥 ), exp10 (10𝑥 ), ln, lg and interpolate (Intrp).

17.3.1

Differential

The differential (d/dt) operator is used to calculate the derivative of the selected source. It is always
used to measure the instantaneous slope of the waveform, such as the slew rate of an operational
amplifier.

int.siglent.com

183

SDS6000L User Manual

The differential equation is:di =y (i + dx) - y(i)dxWhere:
d = Differential result
y = Values of source datai = Data point index
dx = Differential interval
The range of “dx” in the d/dt menu is 4~ number of samples pts.

Note: Differentiation is sensitive to noise. It is helpful to set the acquisition mode
to “Average” or “ERES” to help minimize the visible effects of additional noise.

17.3.2IntegralIntegral operation integrates the waveforms on the screen or within the specified gate.
Setting Offset in the integral menu provides an approach to correct the DC offset of the source.
Small DC offsets in the input signal (or even small offset errors of the oscilloscope itself) may cause
the integral output waveform to "ramp" up or down, as shown below:

Integral without Offset

Integral with Offset

In addition, the integral operation can be performed within a specified gate. Click Gate area in themath dialog box, and enable the Gate function, then set Gate A and Gate B to define the gate.
The setting of the gate cursors is similar to that of normal cursors. See "Select and Move Cursors" fordetails.

184

int.siglent.com

SDS6000L User Manual

17.3.3

Square RootSquare root （√） calculates the square root of the selected source. If the waveform value is negative(the waveform is below the ground level), the result is displayed as zero.

int.siglent.com

185

SDS6000L User Manual

17.3.4Absolute

Absolute (|x|) calculates the absolute value of the selected trace.

17.3.5Sign

In mathematics, the sign function or signum function (from signum, Latin for "sign") is an odd
mathematical function that extracts the sign of a real number.
The sign function of a real number x is defined as follows:Sign(x) = -1

if x < 0,=0

if x = 0,=1

if x > 0.

186

int.siglent.com

SDS6000L User Manual

17.3.6

exp/exp10

The exponential operation includes the exponential operation 𝑒 𝑥 based on constant e and theexponential operation 10𝑥 based on 10.
For example: y(x) = 𝑒 𝑥 .

17.3.7ln/lg

Logarithmic operation includes logarithm base e (ln) and logarithm base 10 (lg). In logarithmic
operation, if the waveform value is negative (the waveform is below the ground level), the result is
displayed as zero.
For example, F1 = 𝑒 𝑥 , where x is the trigonometric wave function. F2 = ln (F1).

int.siglent.com

187

SDS6000L User Manual

17.3.8Interpolate

Between the adjacent sampling points, the waveform is interpolated according to the selectedinterpolation method and interpolation coefficient. Click

Acquire>Interpolation

to set theinterpolation method, and the interpolation coefficient can be set to 2, 5, 10, or 20.

coefficient = 2

coefficient = 20

Filter17.4

The filter operation in the SDS6000L provides FIR (Finite Impulse Response) filtering with the following
types: Low pass, High pass, Bandpass, and Band Reject. The filters have up to 200 taps which canreach roll-off as fast as 0.01*fs. Users just need to set the filter type and corner frequency(s), then the
instrument will automatically calculate the coefficients needed for the filtering.

A.

Choose to let the scope set the sample rate
automatically according to the filter setting or
not. See below for details

B.

Filter type: Low pass, High pass, Bandpass,and Band Reject

C. Corner frequency. For Bandpass and Band
Reject there are two corner frequencies,
separately named Lower Frequency and Upper
Frequency. The attenuation at the cornerfrequency is around 3 dB

188

int.siglent.com

SDS6000L User Manual

Adapt Sample Rate
Unlike an ideal “brick wall” filter, the actual FIR filters have their roll-off from the passband to the
stopband because of the limited number of taps. The 200-tap FIRs applied in the SDS6000L provide
roll-off as fast as 0.01*fs (where fs is the sample rate), this means some limitations in the setting ofcorner frequency at a specified sample rate. The figures below show the limitations in detail:≥0.01fs≥0.01fs≥0.01fs

fc

fc

fs/2

Low pass

fl≥0.01fs≥0.01fsfh

fs/2

High pass≥0.02fs≥0.01fs≥0.01fs

fs/2

Bandpass≥0.01fs≥0.02fsflfh fs/2

Band reject

Take a Low pass filter as an example. It requires that the corner frequency (fc) is not less than 0.01*fs(i.e. fc ≥ 0.01*fs), and the 0.01*fs wide roll-off does not exceed the 1st Nyquist zone (i.e. (fs/2) - fc ≥0.01fs). If fs = 5 GSa/s, then the legal range of the corner frequency (fc) is 50 MHz~2450 MHz
If the expected fc is out of the legal range at some sample rate, then the sample rate needs to bechanged. For instance, when fs = 5 GSa/s, the setting fc = 10 MHz is illegal, so we must lower fs to 1
GSa/s to satisfy the limitation (fc ≥ 0.01*fs).

int.siglent.com

189

SDS6000L User Manual

Professional users who have prior knowledge of FIR filters and are familiar with the operation of this
instrument are encouraged to change the sample rate for the corner frequency of interest. Otherwise,
we recommend using the “Adapt Sample Rate” option. In this mode, the scope forces the Memory
Management Mode to be “Fixed Sample Rate” and automatically adjusts the sample rate to adapt tothe corner frequency setting. For details of the Memory Management Mode refer to the section“Memory Management”.

Note: Be aware that when the corner frequency is set very low the sample rate
may not satisfy the Nyquist Theorem (i.e. fs is no longer greater than or equal
to twice the highest frequency of the input). Keep the sample rate reasonable
at all times.

17.5

Frequency Analysis

The result of FFT (Fast Fourier Transform) calculations is the frequency spectrum of the source signal.
The horizontal axis of the FFT display is labeled using frequency (Hz) units instead of time (seconds).
In addition, the vertical axis provides the option of logarithmic scaling (dBVrms/dBArms or dBm).

190

int.siglent.com

SDS6000L User ManualA.

Time-domain waveform display area

B.

Spectrum (FFT) waveform display area

C. FFT parameter display area
D. Dialog box

Parameter Display Area
The FFT parameters are displayed in the upper right of the spectrum waveform display area:

FFT sample rate (Sa): FFT operation results present the first Nyquist zone (DC ~ Sa/2) of thefrequency spectrum. Be aware that the FFT sample rate may be inconsistent with the sample ratein the time domain. Assuming the max points value is set to 2 Mpts:✓

When the number of points in the time domain, N, is less than 2 Mpts, the FFT takes thenumber which is an integer power of 2 closest to N. In this case, FFT sample rate = sample
rate in the time domain.

✓

When N is greater than 2 Mpts, FFT first decimates N by D and then takes the first 2 Mpts
for the calculation. In this case, FFT sample rate = sample rate in time domain/D.
For example, in the case of sample rate in the time domain is 5 GSa/s and the number ofsamples is 5 Mpts, the FFT first decimates the samples by 2, to 2.5 Mpts, and then takes thefirst 2 Mpts to calculate the spectrum. In this example, FFT sample rate = 5 GSa/s ÷ 2 = 2.5
GSa/s.



FFT points (Curr): The current FFT points, which is an integer power of 2. The SDS6000L
supports up to 8 M points (8388608, to be exact).

Frequency interval ( △ f): The frequency interval between two adjacent points in the FFT
sequence, which is proportional to the frequency resolution.



Average count of FFT (Avg): Displayed only when the FFT mode is set to "Average", indicating
the completed average count.

int.siglent.com

191

SDS6000L User Manual

Select Operation as FFT in the math dialog box andclick Config to recall the configuration dialog box:
A.

Set the maximum points (2n, n = 10 ~ 23)

B.

Set the window type (Rectangle, Blackman,
Hanning, Hamming, and Flattop)

C. Automatically set the horizontal axis to fallspan（0~fs/2）
D. Automatically set the center frequency to the
maximum frequency component in theprevious frameE.

Select the display mode (Split, Full Screen,and Exclusive)F.

Select the FFT mode (Normal, Average, and
Max-Hold)

G. Set the average count in average mode
H. Reset average

Windows
Spectral leakage in FFTs can be considerably decreased when a window is used. SDS6000L provides
five windows that have different characteristics and apply to different scenarios.

For example, for a two-tone signal with a very close frequency interval, it is suitable to use a Rectangle
window with the best frequency resolution. For the case where the accuracy of amplitude measurement
is critical, it is recommended to select the Flattop window with the best amplitude resolution.

192

int.siglent.com

SDS6000L User Manual

Window

Characteristics

Rectangle

The best frequency resolution
The worst amplitude resolution

Main lobe
widthSide lobe
suppression

Maximum
amplitude error

4π/N-13 dB

3.9 dB

8π/N-32 dB

1.4 dB

8π/N-43 dB

1.8 dB

12π/N-58 dB

1.1 dB

23π/N-93 dB< 0.1dB

It is equivalent to the case of
no window
Hanning

Better frequency resolution
Poor amplitude resolution

Hamming

Better frequency resolution
Poor amplitude resolution

Blackman

Poor frequency resolution
Better amplitude resolution

Flattop

The worst frequency resolution
The best amplitude resolution

Display Mode

Split: Time-domain waveform and frequency-domain waveforms are displayed separately. The
time-domain waveform is on the upper half screen, while the frequency-domain waveform islocated within the lower half of the display. In Split mode, if Zoom is enabled, the zoomed waveform
and the frequency domain waveform are displayed on the lower half screen together.



Full Screen: Time-domain waveform and frequency-domain waveform are displayed together.

Exclusive: Only the frequency-domain waveform is displayed.

Split Mode, Zoom off

int.siglent.com

Split Mode, Zoom on

193

SDS6000L User Manual

Full-Screen Mode

Exclusive Mode

FFT ModeNormal: Displays the FFT result of each frame directly.



Max-Hold: Holds the maximum value in the historic frame on the display until cleared. This modeis suitable for detecting discontinuous waves, such as sporadic pulse signals, or frequency
hopping signals. Click Reset in the dialog box to clear the max-hold waveform.



Average: Reduces the effect of the superimposed random noise on the signal. After the FFT modeis set as Average, Average will appear under FFT Mode . The average count can be set
here, ranging from 4 to 1024. Click Reset in the dialog box to restart the average counter.

194

int.siglent.com

SDS6000L User Manual

Click Vertical or Horizontal in the math dialog box torecall the FFT vertical or horizontal settings dialog box:
A.

Set the vertical scale and reference levelB.

Set the unit (dBVrms, Vrms, and dBm). When the unit
is dBm, the oscilloscope will automatically calculate thedBm value based on the Ext Load value set inarea

C. Set the external load, which is used to calculate thecorrect dBm result
D. Return to the previous menuE.

Set the center-span frequencies

F.

Set the start-end frequencies

Unit
The unit of the vertical axis can be set to dBm, dBVrms, or Vrms. dBVrms and Vrms respectively using
either logarithmic or linear scaling. dBVrms is recommended to show larger dynamic ranges. dBm is a
power unit, the correct result can be obtained only if the value of Ext Load is set to be consistent
with the load impedance of the actual measured signal.

Vertical Control


Click Ref Level to set the reference level of the FFT waveform by the mouse wheel or the virtual
keypad.



Click Scale to set the vertical scale of the FFT waveform by the mouse wheel or the virtual
keypad. The reference point for vertical scale scaling is the reference level.

Horizontal Control


Click Center to set the center frequency by the mouse wheel or the virtual keypad.

int.siglent.com

195

SDS6000L User Manual

Click Span to set the frequency span with the center frequency as the center by the mousewheel or the virtual keypad.

Click Start to set the start frequency by the mouse wheel or the virtual keypad.

Click End to set the end frequency by the mouse wheel or the virtual keypad.

The relationship between Center, Span, Start, and End is:Center = (Start + End)/2
Span = End – Start

FFT Tools
The SDS6000L provides two tools for the FFT waveform: Peaks and Markers. The peaks tool can
automatically search the qualified peak points and mark them on the FFT waveform. Up to 10 peaks
are supported. Based on the peak tool, the markers tool can automatically search the qualified
harmonics, and users can control the position of each marker. Up to 8 markers are supported.

Click Tools in the math dialog box to recall the FFT tools dialog box:
When the FFT tool peaks is selected, the dialog box is asfollows:
A.

Turn on or off the table. Turn on the table, the peaks
searched with the limit of

Search Threshold andSearch Excursion will be displayed in a table
B.

Turn on or off the display of peak frequency in the table

C. Sort peaks by amplitude or frequency
D. Set the search threshold. Only peaks larger than thelimit are shownE.

Set the difference between the peak value and theminimum amplitude on both sides. The difference
should be greater than the search excursion

F.

196

Return to the previous menu

int.siglent.com

SDS6000L User Manual

When the FFT tool markers is selected, the dialog box is asfollows:
A.

Control markers. Click to control the display and positionof each marker

B.

Marker on peaks. It will automatically mark the peak thatmeets the conditions of Search Threshold andSearch Excursion

C. Marker on harmonics. It will automatically mark eachharmonic of the FFT waveform
D. Turn on or off the table
E.

Turn on or off the frequency display

F.

Turn on or off the delta display

G. Set the search threshold. Only peaks larger than thepeak limit can be judged as peaks
H. Set the difference between the peak value and theminimum amplitude on both sides. The difference should
be greater than the search excursion, which can be
determined as the peakI.

Return to the previous menu

Note: The FFT dialog box is longer than the display. Slide the dialog box area
up and down by dragging to the view non-displayed areas.

int.siglent.com

197

SDS6000L User Manual

The following screenshot shows the peaks of the FFT waveform:

Measure the FFT waveform
Turn on the cursor function, and then specify the source as "Fx". X1 and X2 cursors can be used tomeasure the frequency value at the cursor position.
Only the “maximum” parameter of the FFT is supported in automatic measurement.

Note: DC components in the signal may show a large amplitude near 0 Hz. If
your application does not require measurement of the DC component, it isrecommended to set the coupling mode of the source channel to "AC".

17.6

Formula Editor

Click Formula Editor in the operation setting page to recall the editor:

198

int.siglent.com

SDS6000L User ManualA.

Formula display text box

B.

Text box operation area, which can clear and modify the entered formula

C. Special operators
D. Operation source. Cx represents analog trace, Zx represents zoomed trace, Fx represents mathtrace and Mx represents memory traceE.

The Keyboard area contains the basic arithmetic operators addition (+), subtraction (-),
multiplication (*), division (/)

F.

Confirm button. After the formula is input, press the button to apply it

The following is an example of F1 = (C1 + C1) * C2 through the formula editor

int.siglent.com

199

SDS6000L User Manual

18 Reference
Data from analog channels or math can be saved to the reference locations (REFA~H) in the built-in
nonvolatile memory. The saved reference waveform can be recalled to be compared with the current
waveform.

A.

Channel waveform

B.

Reference waveform

C. Reference descriptor box
D. The reference dialog box, hidden in this figure

200

int.siglent.com

SDS6000L User Manual

Click + in the descriptor box region and select Ref to recall the ref dialog box.
A.

Select the location of the reference(REFA~REFH)

B.

Select the source (C1~C8 and F1~F4)

C. Enable/disable the Display of thereference waveform
D. Set the label text of the reference trace.
E.

Save the specified waveform in/to thespecified location in

Adjust the Reference Waveform
Refer to "Vertical Setup" for adjusting the reference waveform.

int.siglent.com

201

SDS6000L User Manual

19 Memory
Analog channels (Cx), zoom (Zx), math (Fx), memory waveforms (Mx), and waveforms in files can beimported to available memory waveforms (M1/M2/M3/M4) and displayed on the screen for comparison
with the current waveforms. Differing from reference traces which are displayed data, the memory
traces are in the format of raw data, which can be used as the source of Math and Decoding functions
and can provide more accurate measurements than a reference waveform.

A.

Channel waveform

B.

Memory waveform

C. Memory information
D. The Memory dialog box

There are multiple ways to import waveforms to memory. To enter the memory dialog box:
Click + at the bottom and select M1 ~ M4 .

Click the menu Utility > Save/Recall to open the Save/Recall dialog box, select Mode as"Save”, and Type as "To Memory".

Click the menu Utility > Save/Recall to open the Save/Recall dialog box, select Mode as"Recall", Type as "Waveform", and then import the file to memory.

202

int.siglent.com

SDS6000L User Manual

Details of the dialog box are as below.
A.

Select the location of the memory (M1 ~ M4)

B.

Select the source (Cx / Zx / Fx / Mx / File)

C. Import the trace of the selected source into thememory location set in
D. Turn on or off the MemoryE.

Set the label text of the memory traceF.

Set the vertical parameters of the memory trace

G. Set the horizontal parameters of the memory trace
H. Display the memory information, including source, verticaland horizontal parameters, sampling rate, and length

Adjust the Memory Waveform
There are a couple of ways to adjust the memory waveform:

Adjust the parameters in the descriptor box of Mx at the bottom of the display

Adjust the parameters in the Memory dialog.

In the Horizontal menu, when Sync with Window option is enabled, the horizontal parameters ofthe memory trace and analog traces are adjusted synchronously, otherwise, they are adjusted
separately.

int.siglent.com

203

SDS6000L User Manual

20 Search
The SDS6000L can search for the specified events in a frame. The location of the events is displayedwith white triangle indicators. In YT mode or Roll mode with the acquisition stopped, up to 1250 events
are supported. In Roll mode with the acquisition running, the number of search events is unlimited. The
waveform can be zoomed in when the search function is enabled.

A.

Search event indicator, marking the time point of an event

B.

Total number of events marked on the display

C. The search dialog box, hidden in this example

In the stop state, the areashows the index of the current event/total number of events. The current

event is the one in the center of the display.

204

int.siglent.com

SDS6000L User Manual

Click the menu Analysis > Search to recall thesearch dialog box and turn it on.

Setup Menu
Select and set the search type in the Setup Menu . The SDS6000L provides five search conditions:
Edge, Slope, Pulse, Interval, and Runt.

int.siglent.com

205

SDS6000L User Manual

Search Type

Setup Description

Edge

Slope: Rising, Falling, or Either

Slope

Slope: Rising, Falling
Limit Range setting is available

Pulse

Polarity: Positive, Negative
Limit Range setting is available

IntervalSlope: Rising, Falling
Limit Range setting is available

Runt

Polarity: Positive, Negative
Limit Range setting is available

Search setup is similar to the corresponding trigger type. See the sections “Edge Trigger”, “Slope
Trigger”, ”Pulse Trigger”, ”Interval Trigger” and ”Runt Trigger” for details.

Copy
The SDS6000L supports replication between search settings and trigger settings.
Copy from Trigger: Synchronize the current trigger settings to the search settings.

Copy to Trigger: Synchronize the current search settings to the trigger settings.

Cancel Copy: Cancel the last synchronization and restore the settings before it.

Note: When copying from the trigger, if the trigger type is not supported bysearch, the operation is invalid.

206

int.siglent.com

SDS6000L User Manual

21 Navigate
Click the menu Analysis > Navigate to recall the navigate dialog box. The SDS6000L provides
three navigate types: Search Event, Time, and History Frame.

Navigate by Time
The oscilloscope automatically adjusts the trigger delay
according to the direction set by the user.

Click

Type

in the navigate dialog box to select thenavigate type as “Time”. There are two ways to navigate
by time:

Click the Time area to set the time value by themouse wheel or the virtual keypad.

Click the navigation buttons ◀, ||, or ▶ on the menu toplay backward, stop, or play forward the waveform.
Click the ◀ or ▶ buttons multiple times to speed up theplaying. Three replay speed levels are supported:
Low, Medium , and High Speed.

int.siglent.com

207

SDS6000L User Manual

Navigate by Search Event
When the Search function is turned on and the acquisition is
stopped, Navigate is usable to find search events (see thechapter "Search" for the search function).

Click

Type

in the navigate dialog box to select thenavigate type as “Search Event”.

Click Event Num to set the event number by the mousewheel or the virtual keypad. Click the navigation buttons ◀ or
▶ on the front panel to go to the previous or next search event.

Click Playing Mode to set the search event playing mode.

Click Interval Time to set the playing interval by the mousewheel or the virtual keypad.

Click the Event List Switch area to turn the list on or off. The list contains time labels for each event.
Clicking a row in the list automatically jumps to the corresponding event. This operation is equivalent
to specifying an event in the Event Num area.

208

int.siglent.com

SDS6000L User Manual

Navigate by History Frame
When the history function is turned on, Navigate can be used
to play history frames (see the chapter "History" for details of
the history function).

Click

Type

in the navigate dialog box to select thenavigate type as “History Frame”.

Click Frame to set the frame number by the mouse wheelor the virtual keypad. Click the navigation buttons ◀, ||, or ▶ on
the menu to play backward, stop, or play forward.

Click Playing Mode to set the search event playing mode.

Click Interval Time to set the playing interval by the mousewheel or the virtual keypad.

Click Stop On Search Event to set the stop play condition:
stop when the event is searched. This setting is only valid
when the search function is on.

The following is an example of an occasional runt signal to demonstrate how to quickly locate and findevents of interest by using the combination of Search and Navigate:
The input signal is a 5 V periodic squarewave, in which every 200 ms there is a runt
pulse with a height of 1/3 of the normal
amplitude:

int.siglent.com

209

SDS6000L User ManualFirst, set the trigger type to Runt to trigger on the runt pulse. See the section “Runt Trigger” for details.

Then turn on the Search function and operate Copy from Trigger in the search dialog box so thatthe oscilloscope searches for the dwarf pulses according to the same setting as the trigger.

Set the horizontal scale to 100 ms/div, then 5 markers with ~200 ms interval are shown on the display,
indicating that a total of 5 dwarf pulses were found in the full screen of 1 second time range:

210

int.siglent.com

SDS6000L User Manual

Turn on the Zoom function to observe the full view of the frame and the detail of the third dwarf pulse
at the same time:

Perform Acquire > Run/Stop to stop the acquisition, and then follow the steps
Analysis>

Navigate>Typeto select “Search Event”. The following figure shows the first

dwarf pulse. In this example, the list is enabled and the time labels of each event are shown in the list.

int.siglent.com

2 11

SDS6000L User Manual

22 Mask Test
22.1

Overview

Users can create masks and define the rule used to evaluate Pass/Fail operations. An event violating
the rule is defined as a failure and a pulse can be generated from the "Aux Out" port on the back panel.
This is very useful to find and quantify anomalies in production tests or similar batch measurements.
When Pass/Fail is enabled, the signal output from the Aux Out is automatically switched to Pass/Failpulse.

A.

Mask area in green. Any dot violating the rule is highlighted in red, instead of the normal
waveform color.

B.

Pass/Fail information display area, including the count of the passed frames, failed frames,
total frames, and the fail rate.

C. Dialog box

212

int.siglent.com

SDS6000L User Manual

Perform Analysis > Mask Test to open the Mask Test dialog box:
A.

Turn on/off the test

B.

Select the source（C1~C8）

C. Select the rule（All In, All Out, Any In, and Any Out）
D. Set the mask
E.

Change the state of the Pass/Fail operation. Turning
off the operation when the test is in progress will stop
the test immediately, and the counters in theinformation display area will stop. Turning the test
back on will re-start the test and clear the counters.

F.

Turn on/off the information display

G. Turn on/off “Stop on Fail”. When it is "on", theoscilloscope stops the acquisition once it detects a
failureH. Turn on/off “Failure to History”. When it is "on", thefailure frame will be stored internally and can beviewed through the history function
I.

Turn on/off “Capture on Fail”. When it is "on", thefailure waveform is detected, and the screenshot withthe failure frame will be stored in the external U disk.

J.

Turn on/off the sound prompt when a failure occurs.

int.siglent.com

213

SDS6000L User Manual

22.2

Mask Setup

Click Mask Setup in the Mask Test dialog box to set the mask. There are two methods to create amask, one is by setting horizontal and vertical values, and the other is by drawing a polygon mask.
A.

Create a mask automatically according to
the waveform

B.

Create a custom mask using the Mask
Editor tool

C. Load the mask
D. When “Auto Mask” is On, the mask grows
or shrinks in proportion to the horizontal orvertical setting values. As they change, themask adjusts to fit the new scaling valuesE.

22.2.1Return to the previous menu

Create Mask

The mask can be created based on an existing waveform trace.
A.

Set the spacing of the mask to the trace
horizontal. The units are parts of thehorizontal division setting.

B.

Set the spacing of the mask to the trace
vertical. The units are parts of the verticaldivision setting. Create the mask based on
settings inand

C. Save the mask
D. Return to the previous menu

214

int.siglent.com

SDS6000L User Manual

Set the values for Mask X and Mask Y (in divisions of the display graticule), and then perform
Create Mask to generate the mask. The horizontal and vertical adjustment range is 0.08~4.00 div.

X = 0.2 div, Y = 0.2 div

X = 1 div, Y = 1 div

Saving and recalling mask files (*.msk) is similar to the operation of setup files, see the chapter"Save/Recall" for details.

22.2.2

Mask Editor

The Mask Editor is a built-in tool that provides an approach to create custom masks. Below is its layout:

A.

Menu bar

B.

Coordinate the latest selected point on the display

C. Mask edit area, which is equivalent to the grid area. In this example, a hexagon has beencreated as a part of the mask
int.siglent.com

215

SDS6000L User Manual

D. Toolbar
E.

Coordinate edit area. Set the X ordinate and Y ordinate by the virtual keypad and then click
the “Input” button the perform the ordinate update

F.

Display or hide the coordinates of the polygon vertices on the display

G. Exit the tool

Menu bar
There are 2 menus on the menu bar. The File menu includes ordinary file operations such as:New: Create a new mask file

Open: Open an existing mask file

Save: Save the current mask file

Exit: Exit the Mask Editor tool

The contents of the Edit menu are equivalent to the Toolbar.
Toolbar

Draw: Creates vertices of a polygon by clicking the display or entering values incoordinate edit area

Create Polygon: Creates a polygon based on the drawn vertices by the operation Draw



Undo



Redo



Insert Point: Inserts a vertex on a selected sideEdit Polygon: Edits a polygon. Vertices, sides, and the polygon are all editable object

Delete Polygon: Deletes selected polygon

To edit a vertex, side, or polygon object, first select it, and then move it by mouse or by entering thedesired value incoordinate edit area. For a side, the value is for its middle point. For a polygon,

the ordinate is for its geometric center.

216

int.siglent.com

SDS6000L User Manual

Select a vertex (point)

Select a side (line segment)

Select a polygon

22.3

Pass/Fail Rule

The Pass / Fail rule is specified at the Type region in the Mask Test dialog box.
All In: All data points must be inside the mask to pass the test. Even a single point outside
the mask will cause a failure.



All Out: All data points must be outside the mask to pass the test. Even a single point inside
the mask will cause a failure.



Any In: Any data point inside the mask will be recognized as a pass. All data points outside
the mask will cause a failure.



Any Out: Any data point outside the mask will be recognized as a pass. All data points inside
the mask will cause a failure.

22.4

Operation

Click Operation to start / stop the test. Stopping a test in progress and restarting the test will clear
the count of the passing frames, failed frames, total frames, and the fail rate.

int.siglent.com

217

SDS6000L User Manual

23 DVM
23.1

Overview

The DVM (Digital Voltage Meter) function can be used to measure parameters such as DC and AC
amplitudes. The SDS6000L measures the specified parameter of the input signal and can display it invarious formats, including Bar, Histogram, and Trend. DVM is asynchronous to the acquisition systemof the oscilloscope. Sources of DVM and Measurement can be different, and DVM can work well even
if the acquisition of the oscilloscope is stopped.

A.

Histogram display area

B.

Trend display area

C. Bar display area
D. DVM dialog box

218

int.siglent.com

SDS6000L User Manual

Click the menu Analysis > DVM to open the
DVM dialog box:
A.

Turn on or off the DVM

B.

Select the source (C1~C8)

C. Select the Mode: DC Mean, DC RMS, AC
RMS, Peak-Peak, and Amplitude
D. Turn on or off the Auto Range
E.

Turn on or off the bar diagram

F.

Turn on or off the histogram diagram

G. Turn on or off the trend plot
H. Turn on or off Hold. DVM stops
acquisition in Hold modeI.

Turn on the overload alarm, an alarm willsound when the signal is out of range

23.2

Mode

DVM provides 5 modes. Click mode in the DVM dialog box to open the mode selection window:

int.siglent.com

219

SDS6000L User Manual

DC Mean: Average of the data

DC RMS: Root mean square of the data at DC coupling

AC RMS: Root mean square of the data at AC coupling

Peak-Peak: Difference between maximum and minimum data valueAmplitude: Difference between top and base in a bimodal waveform. If not bimodal, the difference
between max and min.

23.3

Diagrams

After selecting the mode, users can click the screen to open the state diagrams: bar, histogram, andtrend. The color of the data in the diagrams is consistent with the color of the source.

Bar
The bar diagram accurately displays the current measured value. Click Bar in the DVM dialog box
to display it.

A.

Mode

B.

Current value

C. Bar corresponding to the current value

Histogram
A histogram visually indicates the probability distribution of the measured values. Click histogram in
the DVM dialog box to display it.

220

int.siglent.com

SDS6000L User Manual

A.

Mode

B.

Histogram display area

C. Current value
D. The bin includes the maximum value and probability with which values fall into it
E.

Statistics counts

F.

Reset statistics

Trend
The trend diagram indicates the trend of the measured values over time. Click trend in the DVM
dialog box to display it.

int.siglent.com

221

SDS6000L User Manual

A.

Mode

B.

Trend display area

C. Extend the range of time. Click it to expand the time range.
D. Current valueE.

Maximum value

F.

Minimum value

G. Average value
H. Reset statistics

Click the symbolin the histogram and trend display areas to restart statistics.

When all the 3 diagrams are closed, there is a simple information bar on the top-left of the waveform
display area to show the current value of the DVM:

222

int.siglent.com

SDS6000L User Manual

24 Counter
Overview

24.1

The counter is used to measure the frequency and period of a signal or count the events happening
within it. The counter is asynchronous to the acquisition system of the oscilloscope. It can work welleven if the acquisition of the oscilloscope is stopped.

Click the menu Analysis > Counter to open the counter dialog box:
A.

Turn on or off Counter

B.

Select the mode: frequency,
period, and totalizer

C. Select the source (C1~C8)
D. Set the level of the counter
E.

Turn on or off statistics

F.

Clear and restart the statistics.

Mode
See the section “Mode” for details.

Statistics
When statistics are enabled, the counter will increment the data and display the statistics results on
the screen.

int.siglent.com

223

SDS6000L User Manual

Value – The latest count
Mean – The average of all historical counts
Min – The minimum of all historical counts
Max – The maximum of all historical counts
Stdev – The standard deviation of all historical counts, used to judge the distribution of historical countparameters
Count – Number of counts obtained
Level – Counter level

Click Reset Statistics in the measure dialog box to clear and restart statistics.

224

int.siglent.com

SDS6000L User Manual

Mode

24.2

The counter provides 3 modes. Click mode in the counter dialog box to open the mode selection
window:Frequency: Average frequency over a set period

Period: The reciprocal of the average frequency over a set period

Totalizer: Cumulative countWhen the mode is set to totalizer, the edge of the counting source needs to be set. When the counting
gate is turned on, the counter can count only when the gate-source meets the specified conditions.
Click Gate Setting to recall the gate setting dialog box:
A.

Turn on or off the gate

B.

Select the gate type: Level or After Edge

C. Gate source display area. C1 and C2 are gate sources ofeach other, C3 and C4 are gate sources of each other, C5
and C6 are gate sources of each other, C7 and C8 are gate
sources of each other
D. When the gate type is level, set the polarity (Positive or
Negative) of the gate source. When the gate type is edge,
set the slope (Rising or Falling edge) of the gate source.
E.

Set the gate level

F.

Return to the previous menu

int.siglent.com

225

SDS6000L User Manual

25 Histogram
Overview

25.1

The SDS6000L supports waveform histograms for observing probability distributions of the waveformin the specified region. The statistics can be performed in both horizontal and vertical directions. The
histogram continues to update as long as the acquisition is active.

Click Analysis > Histogram to recall the Histogram dialog box:
A.

Turn on or off the histogram

B.

Select the source of the histogram:
C1~C8. When Zoom is turned on, thesource selection is automaticallyswitched to Z1-Z8

C. Set the type of histogram: Horizontal,
Vertical or Both
D. Set the region of the histogram. The
setting range is within the waveform
area.
E.

Turn on or off histogram statistics

F.

Clear and restart the histogram
statistics.

TypeHorizontal – Displays the histogram in the horizontal direction. The oscilloscope counts waveform
data falling into every horizontal (time) bin defined by the histogram region and shows thehistogram at the bottom of the waveform area.



Vertical – Displays histogram in the vertical direction. The oscilloscope counts waveform data
falling into every vertical (amplitude) bin defined by the histogram region and shows the histogramat the left of the waveform area.



226

Both – Displays both horizontal and vertical histograms

int.siglent.com

SDS6000L User Manual

Horizontal

Vertical

BothRegion Setting
See the section “Region Setting” for details.

Histogram Statistics
When histogram statistics is turned on, the oscilloscope will show the statistical parameters of thehistogram on the display.

int.siglent.com

227

SDS6000L User Manual

Sum – Total samples falling in the histogram region
Peak – Samples in the highest bin
Max – Maximum value of the samples
Min – Minimum value of the samples
Pk-Pk – Maximum - Minimum
Mean – Mathematical expectation (or average) value of the samples
Median – The value separating the higher half from the lower half of the histogram
Mode – The value that appears most often
Bin Width – The width of each bin
Sigma – Standard deviation of the samples
Click Reset Histogram in the histogram dialog box to clear and restart the statistics.

25.2

Region Setting

Histogram region can be created and moved directly by the mouse or by setting Histogram >Region Setting in the dialog box. The border color of the histogram region is consistent with the colorof the specified source.
By the mouse
Click any position of the waveform area and draw a rectangular box, as follow:

228

int.siglent.com

SDS6000L User Manual

A menu pops up at the end of the drawing action. Select "Histogram" in the menu:

After the region is created, it can be moved by dragging action.

Dialog Box
Click Histogram > Region Setting to recall the dialog box.
A.

Set the left border of the histogram region

B.

Set the right border of the histogram region

C. Set the top border of the histogram region
D. Set the bottom border of the histogram region
E.

Return to the previous menu

Click the area above to set the value by the mouse wheelor the virtual keypad.

int.siglent.com

229

SDS6000L User Manual

26 Power Analysis
26.1

Overview

The SDS6000L supports a power analysis function. Power analysis can help users quickly and easily
analyze and debug switching power supply design. It automatically calculates Power Quality, Current
Harmonics, Inrush Current, Switching Loss, Slew Rate, Modulation, Turn On/Turn Off, Transient
Response, PSRR, Power Efficiency, Output Ripple, etc. Full use of the Power analysis requires adifferential voltage probe like the SIGLENT DPB series, a current probe like the SIGLENT CP series,
the SIGLENT DF2001A deskew fixture, and the power analysis activation license (part number
SDS6000L-PA). Install the software option part number SDS6000L-PA to permanently enable thepower analysis function.
Click Analysis > Power Analysis to recall the power analysis dialog box:
A.

Enable/Disable the test

B.

Select the analysis item (Power Quality,
Current Harmonics, Inrush Current, Switching
Loss, Slew Rate, Modulation, Turn On/Turn
Off, Transient Response, PSRR, Power
Efficiency, Output Ripple, etc.)

C. Turn on/off the state of the analysis item

26.2

Power Quality

The specific measurement parameters of power quality analysis include active power, apparent power,
reactive power, power factor, power phase angle, voltage effective value, current effective value,
voltage crest factor, and current crest factor of power input of a switching power supply.

TypePower -- Includes all the items to describe energy flow in a system: Active power, reactive power,
apparent power, power factor, and power phase angle.

230

int.siglent.com

SDS6000L User Manual

1

P: Active Power = 𝑁 ∗ ∑𝑁−1
𝑖=0 𝑉𝑖 ∗ 𝐼𝑖
S: Apparent Power = Vrms * Irms
Q: Reactive Power =√Apparent Power2 − Active Power 2
φ: Power Phase Angle: Phase difference between voltage and currentcos φ: Power Factor, which is the ratio of active power and apparent power.

Voltage Crest -- Voltage parameters of the power input include voltage crest, voltage effective value,and voltage crest factor.
2
Vrms = 1 ∗ √∑𝑁−1
𝑖=0 𝑉𝑖𝑁

V_Crest = Vpeak / Vrms

Current Crest -- Current parameters of the power input include current crest, current effective value,and current crest factor.
2
Irms = 1 ∗ √∑𝑁−1
𝑖=0 𝐼𝑖𝑁

I_Crest = Ipeak / Irms

int.siglent.com

231

SDS6000L User Manual

Signal Settings
Click Signal to recall signal settings dialog box:
A.

Set the input voltage source

B.

Set the input current source

C. Set the periods displayed
D. View the connection guideE.

Return to the previous menu

Connection Guide
Click Signal > Connection Guide to recall the connection guide of power quality, as shown in the
figure below. Please follow the instructions in this figure for correct connections. Click the icon on the
top right of the guide to close.

232

int.siglent.com

SDS6000L User Manual

26.3

Current Harmonics

Current harmonics are used to analyze the input current harmonics. An FFT of the selected channel isperformed to get the harmonic components. The signal settings and connection guide are the same as
the power quality test.

Configuration
Click Config to recall the configuration dialog box:
A.

Set the frequency of the line (Auto, 50 Hz,
60 Hz, or 400 Hz)

B.

Set the standard type (IEC61000-3-2 A,
IEC61000-3-2 B, IEC61000-3-2 C or
IEC61000-3-2 D)

C. Set the display type (Off, Bar Chart, or Table)
D. Return to the previous menu

Standard
IEC 61000-3-2 is an international standard that limits mains voltage distortion by prescribing the
maximum value for harmonic currents from the second harmonic up to and including the 40th harmonic
current. There are 4 different classes in the EN 61000-3-2 that have different limit values:Class A: Balanced 3-phase equipment, household appliances excluding equipment identified asclass D, tools, excluding portable tools, dimmers for incandescent lamps, audio equipment, andall other equipment, except that stated in one of the following classes.

Class B: Portable tools, arc welding equipment which is not professional equipment

Class C: Lighting equipment.

Class D: PC, PC monitors, radio, or TV receivers. Input power P ≤ 600 W.I

int.siglent.com

233

SDS6000L User Manual

Parameter Description
For the first 40 harmonics, the following values are displayed:

Measured Value (RMS) -- The measured value displayed in the unit specified by the harmonic unit
parameter
Limit Value (RMS) -- Limits specified by the selected standard
Margin (%) --The margin specified by selected standard parameters. The margin value is (standard
value - measured value) / standard value * 100%
Pass/Fail State -- Determine whether the measured value is passed or failed according to the selected
standard. The rows in the table or the bars in the bar chart are rendered in different colors based on
the pass / fail state. When the value is greater than 85% of the limit but less than 100% of the limit, it
is defined as a critical state.
Total Harmonic Distortion（THD）=100% ∗√X22 +X23 +⋯+X2𝑛， where Xn is the nth harmonic, and X1 is

X1

the basic component.

234

int.siglent.com

SDS6000L User Manual

26.4

Inrush Current

A large current far greater than the stable current may flow through at the moment of switching on a
power supply. The large current is called the inrush current. The current waveform when switching on
is shown in the figure below:

Signal Settings
Click Signal to recall signal settings dialog box:
A.

Set the input voltage source

B.

Set the input current source

C. Set the expected current value, the range is 100 mA~ 500 A. The oscilloscope will set the trigger level toan expected current/20 and the vertical scale of thecurrent channel to an expected current/3.
D. Set the maximum effective input voltage (Vrms), therange is 1 V ~ 1 kV. The oscilloscope will set thevertical scale of the voltage channel to maximum
input voltage/6
E.

View the connection guide

F.

Return to the previous menu

int.siglent.com

235

SDS6000L User Manual

Switching Loss

26.5

Switch loss analysis can be used to calculate the power dissipated in the switching period.

Deskew Calibration
A relatively small skew can cause a large measurement error of switching loss, especially during theon phase when the voltage is close to zero and the non-on phase when the current is close to zero.
This is a typical oscilloscope dynamic range limitation when trying to measure weak voltage and (or)current in the presence of relatively large switching voltage and (or) current.

To correct the skew between the oscilloscope channels or probes, the deskew procedure should be
performed once initially, and re-run when any part of the hardware setup changes (for example, adifferent probe, different oscilloscope channel, etc.) or when the ambient temperature changes. The
deskew procedures by using the DF2001A deskew fixture are as follows:A.

Demagnetize and zero-adjust the current probe

B.

Set up connections to the DF2001A deskew fixture

Available current
probesSmall Loop

Large Loop

CP4050 (1 MHz,50 Arms)

CP5150 (12 MHz,150 Arms)

CP4070A (300 kHz,70 Arms)

CP5500 (5 MHz,500 Arms)

CP5030 (50 MHz,30 Arms)
CP5030 (100 MHz,30 Arms)
High-voltage
differential probe
sense points

J7: signal terminal

J3: signal terminal

J6: earth terminal

J4: earth terminala)Connect D+ and D- of the high-voltage differential probe to J7 and J6b)Connect the current probe to the loop with the direction of the arrow indicating the directionof the current flowc)

Make sure the switch S1 on the deskew fixture is set to the appropriate side of the fixture(either “small loop” or “large loop”)

d)Connect the deskew fixture to a USB port on your oscilloscope or a PC using a USB cable.
The USB port supplies power to the deskew fixture

236

int.siglent.com

SDS6000L User Manual

C. Connect the voltage and current probes to the oscilloscope input channel
D. Click Signal to enter the signal setting menu. Select the corresponding input channel, andthen click

Auto Deskew

to perform the calibration. When the process completes, amessage will pop up indicating whether the deskew operation was successful or not.

Before deskew calibration

int.siglent.com

After deskew calibration

237

SDS6000L User Manual

Configuration
Click Config to recall the configuration dialog box:
A.

Set the voltage reference, i.e. the switch level at theedge of the input switch. This value is a percentage ofthe maximum switching voltage. Adjusting the value
to ignore background noise. This value is used todetermine the threshold hysteresis of the switch edge

B.

Set the current reference, i.e. the switch level at the
beginning of the input switch edge. This value is thepercentage of the maximum switching current. Thisvalue can be adjusted to ignore background noise orinvalid offset that is difficult to eliminate in the current
probe. This value is used to determine the threshold
hysteresis of the switch edge

C. Set the conduction type (Waveform, Rds(on) orVce(sat))D. Set Rds(on) resistanceE.

Set Vce voltage

F.

Return to the previous menu

Conduction Type
Waveform -- The power waveform uses the original data, and the calculation formula is P = V * I, E =P * T.
Rds (on) -- In the on area (where the voltage level is lower than the voltage reference (adjustable)),the power calculation formula is P = I2 * Rds (on). In the off area (where the current level is lower thanthe current reference (adjustable)), the power calculation formula is P = 0 watt.
Vce (sat) -- In the on area (where the voltage level is lower than the voltage reference (adjustable)),the power calculation formula is P = Vce (sat) * I. In the off area (where the current level is lower thanthe current reference (adjustable)), the power calculation formula is P = 0 watt.

238

int.siglent.com

SDS6000L User Manual

Connection Guide
Click Signal > Connection Guide to recall the connection guide of switching loss, as shown in
the figure below. Please follow the instructions in this figure for connection. Click the icon on the top
right of the guide to close.

26.6

Slew Rate

The slew rate measures the change rate of voltage or current during switching.

Connection Guide
Click Signal > Connection Guide to recall the connection guide of the slew rate, as shown in the
figure below. Please follow the instructions in this figure for connection. Click the icon on the top rightof the guide to close.

int.siglent.com

239

SDS6000L User Manual

26.7

Modulation

Modulation analysis measures the control pulse signal of the switching device (MOSFET) and observes
the pulse width, duty, period, frequency, and other trends of the control pulse signal in response todifferent events.
Connection Guide
Click Signal > Connection Guide to recall the connection guide of modulation, as shown in the
figure below. Please follow the instructions in this figure for connection. Click the icon on the top rightof the guide to close.

26.8

Output Ripple

Power supply ripple is an important parameter to evaluate DC power supply, which represents thequality of output DC voltage. Ripple analysis can measure the current value, average value, minimum
value, maximum value, standard deviation, and count of the power supply output ripple.
Connection Guide
Click Signal > Connection Guide to recall the connection guide of output ripple, as shown in the
figure below. Please follow the instructions in this figure for connection. Click the icon on the top rightof the guide to close.

240

int.siglent.com

SDS6000L User Manual

26.9

Turn On/Turn Off

Turn-on analysis determines the time taken for the power supply to reach 90% of its steady-state output.
Turn-off analysis determines the time taken for the power supply to fall to 10% of its maximum output
voltage.

Signal Settings
Click Signal to recall signal settings dialog box:
A.

Set the input voltage source

B.

Set the output voltage source

C. Set the input type (AC or DC)
D. Set the duration, the range is 5 ns ~ 20 s. The
oscilloscope will set the timebase according to
the value (duration / 10)
E.

Set the maximum effective input voltage Vrms,
the range is 1 V ~ 1 kV. The oscilloscope will
set the vertical scale of the voltage channel tomaximum input voltage / 7 and the trigger levelto maximum input voltage / 10.

F.

Set the stable output voltage value, the range is-30 V ~ 30 V. The oscilloscope will set thevertical scale of the output voltage channel tostable output voltage / 6

G. View the connection guide
H. Return to the previous menu

Testing Conditions
Turn On -- Determines the time taken for the power to reach a certain percentage of its steady-state
output. The turn on time is between T2 and T1, where:
T1 = When the input voltage first rises to a certain percentage (usually 10%) of its maximum amplitude
T2 = When the output DC voltage rises to a certain percentage (usually 90%) of its maximum amplitude

int.siglent.com

241

SDS6000L User Manual

Turn Off -- Determines the time it has taken for the power supply to be turned off to fall to a certain
percentage of its maximum output voltage. The turn off time is between T2 and T1, where:
T1 = When the input voltage finally falls to a certain percentage (usually 10%) of its maximum amplitude
T2 = When the output DC voltage finally falls to a certain percentage (usually 10%) of its maximum
amplitude

Connection Guide
Click Signal > Connection Guide to recall the connection guide of turn on/turn off, as shown in
the figure below. Please follow the instructions in this figure for connection. Click the icon on the top
right of the guide to close.

26.10 Transient Response
Transient response analysis can determine the response speed of the output voltage of the power
supply to the change of the output load. This time starts from the first time that the output voltage exits
the stable band and ends at the last time that the output voltage enters the stable band.

Signal Settings
Click Signal to recall signal settings dialog box:

242

int.siglent.com

SDS6000L User ManualA.

Set the output voltage source

B.

Set the output current source

C. Set the duration. The oscilloscope will set theappropriate time base according to theduration
D. Set the stable output voltage, i.e. the expected
output DC voltage of the power supply in astable stateE.

Set the overshoot percentage of output
voltage, which can be used to determine thestable band value of transient response andadjust the vertical scale of the oscilloscopeF.

View the connection guide

G. Return to the previous menu

Configuration
Click Config to recall the configuration dialog box:
A.

Set the input low current value. This is the
expected low current value before or after
the load change

B.

Set the input high current value. This is the
expected high current value before or after
the load change

C. Return to the previous menu

Note: The low and high current values are used to calculate the trigger level
and adjust the oscilloscope vertical scale. After the load changes, the current
value changes from low to high (or from high to low), and the oscilloscope
triggers and performs to measure the transient response stability time.

int.siglent.com

243

SDS6000L User Manual

Connection Guide
Click Signal > Connection Guide to recall the connection guide of transient response, as shownin the figure below. Please follow the instructions in this figure for connection. Click the icon on the top
right of the guide to close.

26.11 PSRR
The power supply rejection ratio (PSRR) test is used to determine how the regulator suppresses ripple
noise in different frequency ranges. The oscilloscope controls the arbitrary waveform/function
generator to output a sweep signal, which is used to introduce ripple into the DC voltage transmitted
to the voltage regulator. Measure the AC RMS ratio of input to output, and plot the relationship between
the ratio and frequency. The background noise of the oscilloscope is higher than that of the network
analyzer, and the sensitivity is lower than that of the network analyzer, so the PSRR measured by theoscilloscope is difficult to exceed -60 dB. The PSRR test is generally acceptable for sampling theoverall PSRR behavior of the power supply under test.
The configuration of PSRR is the same as that of the Bode plot, see the chapter "Bode Plot" for details.

Connection Guide
Click Signal > Connection Guide to recall the connection guide of PSRR, as shown in the figure
below. Please follow the instructions in this figure for connection. Click the icon on the top right of theguide to close.

244

int.siglent.com

SDS6000L User Manual

26.12 Power Efficiency
Power efficiency analysis can test the overall efficiency of the power supply by measuring the output
power and input power.

Connection Guide
Click Signal > Connection Guide to recall the connection guide of power efficiency, as shown in
the figure below. Please follow the instructions in this figure for connection. Click the icon on the top
right of the guide to close.

26.13 SOA
The safe operating area (SOA) of MOSFETs defines the voltage, current, and power consumption
conditions over which the device can be expected to operate without self-damage. The oscilloscope
can automatically generate the SOA according to the Voltage Limit , Current Limit , andPower Limitparameters set in the

Configmenu, and judge if the stress on the MOSFET isbeyond the SOA or not. This helps designers quickly find a problem or latent risk in the circuit.

Connection Guide
Click Signal > Connection Guide to recall the connection guide of power efficiency, as shown in
the figure below. Please follow the instructions in this figure for connection. Click the icon on the top
right of the guide to close.

int.siglent.com

245

SDS6000L User Manual

Note: Perform a Deskew operation between the voltage and current input
channels before the test. See the section “Switching Loss” for more details
on deskewing.

The test steps are as followed:
1.

In the Input Setup menu, assign the correct channels to Input Voltage and Input Current .
Set Duration as the expected observed duration, then the oscilloscope will automatically set
the timebase according to it.

2.

In theConfigmenu set Voltage Limit, Current Limit, andPower Limitparameters

according to the datasheet of the MOSFET device under test, then the oscilloscope willautomatically generate the SOA and the vertical scale of the Input Voltage and Input Current
channels according to them.
3.

Turn on Test State . The oscilloscope will begin to acquire the voltage and current waveforms,
and display the SOA measurement, showing if the measured stresses are in the SOA or not.

4.

246

Users can adjust horizontal, vertical, and trigger settings during the test for the best observation.

int.siglent.com

SDS6000L User Manual

Below is an example of testing the power-up stresses on a MOSFET and judging if the stresses aresafe for the MOSFET using the SOA:

Voltage and current waveforms on the MOSFET at power-up

SOA, in which the mask is created according to the voltage limit, current limit, and power
limit. The result shows “Pass” because all the stresses are within the mask.

int.siglent.com

247

SDS6000L User Manual

27 Bode Plot
27.1

Overview

The SDS6000L supports an automatic Bode plot function. This function provides a frequency response
curve of the device-under-test as well as the interface for output sweep parameter control and data
display settings. At this time, either the built-in waveform generator or one of the SIGLENT SDG series
arbitrary function generators are supported. During the sweep, the oscilloscope configures thegenerator output frequency and amplitude and then compares the input signal to the output of the DUT.
Gain (G) and phase (P) are measured at each frequency and plotted on the frequency response Bodeplot. When the loop response analysis is complete, you can move the markers on the chart to see thegain and phase values measured at each frequency point. You can also adjust the scale and offset
settings for the amplitude and phase plots.

Click Analysis > Bode Plot to recall the Bode plot dialog box:
A.

Turn on/off the Bode plot

B.

Configure the Bode plot (DUT, AWG connection,
sweep parameters)

C. Turn on/off the operation
D. Set display parameters, including coordinate axis,
trace visibility, and cursors
E.

Data list. Open the data list of the Bode plot toview the curve data, save the data results to a
U-disk or recall it from a U-diskF.

Set the measurement parameters. Parametermeasurement of the scanning curve includes
upper cut-off frequency (UF), lower cut-off
frequency (LF), bandwidth (BW), gain margin(GM), and phase margin (PM)G. Quickly print the specified Bode Plot waveformarea to the storage.

248

int.siglent.com

SDS6000L User Manual

27.2

Configuration

27.2.1ConnectionA.

DUT input and output channels

B.

Channel gain. When it is set to Auto, the oscilloscope will automatically adapt the verticalscale according to the signal amplitude; when it is set to Hold, it will always keep the verticalscale before the test operation

C. Arbitrary waveform generator connection settings. Click Interface to select the connection
type. When selecting LAN, it needs to

Set IP

andSave

it. Click

Test

to confirm

whether the arbitrary waveform generator is connected correctly.

27.2.2

Sweep

Click Sweep Type to select the sweep type. There are two types: Simple and Variable level.

A.

Set the sweep type

B.

Set the sweep as Continue or Single

int.siglent.com

249

SDS6000L User Manual

C. Set the sweep frequency. The frequency mode can be linear or decade. When it is set tolinear, the corresponding center frequency and span frequency need to be set; when it is set
to a decade, the corresponding start frequency and stop frequency need to be set
D. Set the number of sweep points. The larger the number of points, the higher the sweep
resolutionE.

Set the amplitude of the sweep signal

F.

Set the offset of the sweep signal

G. Set the signal amplitude unit. When it is set to dB, the reference level and load need to be
set
H. Set the loadA.

Set the sweep type to Vari-Level

B.

Select a profile. Up to 4 profiles can be edited

C. Set the frequency mode
D. Set the number of sweep points
E.

Set the offset of the sweep signal

F.

Set the signal amplitude unit. When it is set to dB, the reference level and load need to beset

G. Variable level signal display area
H. Click Edit to turn on the profile editor

250

int.siglent.com

SDS6000L User Manual

Click Nodes to set the signal node number through the mouse wheel, or click ▲ to increase thenodes and ▼ to decrease.

Click the cell in the table area to set the frequency and amplitude of the corresponding node. Click toactivate the cell, adjust the value through the mouse wheel, or click the cell again to call up the virtual
keypad for setting.

int.siglent.com

251

SDS6000L User Manual

27.3

Display

Bode plot display settings include amplitude, phase, cursors, and trace visibility.

Amplitude
Set the amplitude coordinate axis of the Bode plot. Click

Display>

Amplitude

to recall theamplitude setting dialog box:
A.

Set the reference position to the center or top

B.

Set the scale of the amplitude coordinate axis

C. Set the reference level, the max value of theamplitude coordinate axis
D. Set the amplitude mode. Set to Vout to display
the amplitude value of the output signal; set to
Vout/Vin to display the amplitude ratio of theoutput signal to the input signal
E.

When the mode is Vout, it needs to set theunit (Vpp, Vrms, dBV, dBu, dBm, or ArbitrarydB); when the mode is Vout/Vin, it needs to
set the amplitude axis type (Linear or
Logarithmic).

F.

Auto-Set. According to the output signal
amplitude curve, the oscilloscope
automatically sets the scale and reference
level

G. Return to the previous menu

Phase
Set the phase coordinate axis of the Bode plot. Click Display > Phase to recall the phase setting
dialog box:

252

int.siglent.com

SDS6000L User ManualA.

Set the reference position to the center or top

B.

Set the scale of the phase coordinate axis

C. Set the reference level, the max value of thephase coordinate axis
D. Set the phase unit (Degree or Rad)E.

Auto-Set. According to the output signal phasecurve, the oscilloscope automatically sets thescale and reference level

F.

Return to the previous menu

Cursors
SDS6000L can use the cursor to measure the Bode plot curve. The cursor of the Bode plot is similar
to the ordinary cursor, see the chapter "Cursors" for details. Click Display > Cursors to recall the
cursor setting dialog box.
A.

Turn on/off the cursors of the Bode plotB.

Set the cursors type (X or Y)

C. Specify the cursor and set the position
D. Set the source 1
E.

Set the source 2

F.

Return to the previous menu

int.siglent.com

253

SDS6000L User ManualTrace Visibility
When multiple output signals are connected, the Bode plot interface will display the amplitude andphase curves of all output signals at the same time. Users can turn on/off other scanning curves toobserve the details of specific curves. Click Display > Trace Visibility to recall the setting dialog
box.

27.4

Data Analysis

With the data list, cursor measurement, and automatic measurement functions, the Bode plot curve
can be analyzed in detail. The data list provides the raw values for each point. The cursors can beused to flexibly measure the change of each position of the curve. The automatic measurement function
can be used to measure the five parameters of the Bode plot curve: upper cut-off frequency (UF), lower
cut-off frequency (LF), bandwidth (BW), gain margin (GM), and phase margin (PM).

A.

Data list display area

B.

Cursors

C. Data point location line
D. Cursor information display area

254

E.

Measurement parameters display area

F.

Bode plot dialog box

int.siglent.com

SDS6000L User Manual

Data List
Click Data to recall the data setting dialog box:
A.

Turn on/off the data list

B.

Set the selected row in the list. Adjust
the mouse wheel to set, or directly
click the list display area to select a
specific line.

C. Set the data source
D. Save/Recall the data. Saving andrecalling the Bode plot data (*.csv) issimilar to the operation of setup files,
see the chapter "Save/Recall" fordetails
E.

Return to the previous menu

MeasureClick Measure to recall the measure setting dialog box:

int.siglent.com

255

SDS6000L User ManualA.

Set the position of measurement items, and support 5 measurement items at mostB.

Set the measurement source

C. Measurement parameter area. Click each parameter area to activate the measurement
parameter. In the figure above, "UF" is activated.
D. Clear all measurements
E.

Clear the current measurement

F.

Close the measurement configuration window

For example, to add UF measurements for C2 and GM measurements for C3, follow the steps below:
Measure > Position > Source > UF
Measure > Position > Source > GM

256

int.siglent.com

SDS6000L User Manual

28 Eye Diagram
28.1

Overview

An eye diagram is an oscilloscope display in which a high-speed digital signal is repetitively sampled
and applied to the vertical input, while the data rate is used to trigger the horizontal sweep. It is called
an Eye because the pattern looks like a series of eyes between a pair of rails. The combined effects ofchannel and dispersion can be evaluated from the Eye opening width and height.

Click Analysis > Eye Diagram to recall the diagram analysis dialog box, this process may takeseveral seconds. The user interface is similar to normal mode.

A.

Menu Bar

B.

Grid Area

C. Dialog Box
D. Measurement parameters display areaE.

Eye diagram descriptor box, click to recall the eye diagram dialog box

F.

Timebase and Trigger descriptor box

int.siglent.com

257

SDS6000L User Manual

Eye Diagram Dialog Box:A.

Turn on/off the eye diagram

B.

Quick view. The oscilloscope will automaticallyadjust its vertical/horizontal scale and trigger
level after the corresponding channel source is
set and the quick view option is active

C. The signal setting, including source selection
and level adjustment
D. Clock recovery options, including ConstantFrequency and PLL
E.

Measure, similar to Measure in normal modeF.

Mask test

G. Turn on / off the operation. It is allowed toenter the Y-T mode without exiting the eye
diagram if run off the operation.
H. Run / Stop eye diagram when the operation is
active.

258

int.siglent.com

SDS6000L User Manual

Signal Setting

28.2

Click Signal Setting to open the signal setup dialog box.
A.

Signal Type: NRZ, 100Base-TX

B.

The source can be C1~C8.

C. Set the trigger level manually
D. The hysteresis of the trigger level has a unitof division in the vertical direction. A Larger
hysteresis range comes with better noise
tolerance but reduced sensitivity. A larger
hysteresis range requires a higher signal
amplitude.
E.

The find level indicates that the oscilloscope
will automatically seek and set the trigger
level at the intersection of the rising andfalling edges of the eye diagram.

Table 28.1 Description of Signal Type
Signal Type

Description

NRZ

General NRZ (non-return-to-zero) code

100Base-TX

MLT-3 (Multi-Level Transmit) code that uses three voltage
levels. When Signal Type = 100Base-TX, there are twoavailable levels (upper level for the upper eyes, and lower
level for the lower eyes);

int.siglent.com

259

SDS6000L User Manual

28.3

Clock Recovery

Clock recovery is a feature that provides an ideal clock for comparison to actual signal edges. It is the
key step in constructing an eye. The eye diagram can be constructed only when the reference clock isrecovered from the data correctly. Click Clock Recovery in the eye diagram menu to open the dialog
box.
A.

Set the clock recovery method, ConstantFrequency, or PLL. See Table 28.2 fordetails.

B.

Choose the mode of finding rate between
Manual and Automatic.

C. Choose the find rate between Every andFirst. The oscilloscope will perform aclock recovery for each acquisition in the“Every” mode and at each new frame, theeye diagram will replace the clock ratewith the one from the previous frame. In
the “First” mode, the oscilloscope uses
the clock recovered from the first frame
for all the frames

Table 28.2 Description of Clock Recovery Method
Clock Recovery

Description

MethodConstant

The clock is assumed to be a constant frequency and recovered

Frequency

through the linear fitting based on the least-squares method.

Phase-Locked

The value of each reference clock edge will be calculated by the

Loop

PLL software based on the sampled signal. Using this method, thelow-frequency components in the jitter can be removed because of
the tracking capacity of the PLL. This method supports First Order
Golden PLL and its loop bandwidth can be configured by changing
the damping factor.

260

int.siglent.com

SDS6000L User Manual

Table 28.3 Description of Clock Data Rate
Date Rate

Description

Automatic

The clock is recovered using the signal edge database only. It is
the recommended method in most cases.

ManualThe oscilloscope will perform clock recovery under the expected
Data Rate as your input. Suitable for the situations in which theoscilloscope cannot recover the clock accurately when the signalto-noise ratio is poor.

28.4

Measurement

Click Measure under the eye diagram menu to open the measurement setting dialog.
The measurement under the eye diagram mode is similar to measurements in normal mode. Refer to
Chapter “Measurement” to see the operating instructions.

Table 28.4 Description of Eye Diagram Parameters
Parameters

Description

Time Interval

TIE is the difference between the edges in a data signal and the edges inError

an ideal data signal

Eye Width

The horizontal opening of an eye diagram. Estimated based on the
distribution of the eye diagram crossing point in the horizontal direction.

EyeWidth= ( t crossin g 2 -3crossin g 2 ) - ( t crossin g1 +3crossin g1 )int.siglent.com

261

SDS6000L User Manual

Eye Height

The vertical opening of an eye diagram. Estimated based on the vertical
distribution of the measured high-voltage and measured low-voltage in the
40% to 60% UI interval.

EyeHeight= ( V1 -31 ) - ( V0 +30 )1-levelThe high voltage corresponds to 1-level in the diagram. The one level iscalculated as the mean value of the top histogram distribution in the middle
20 percent of the eye period

0-levelThe low voltage corresponds to the 0-level in the diagram. The zero level
comes from the histogram mean value of the data captured inside themiddle 20 percent of the eye period

Eye Amplitude

The eye amplitude describes the voltage between 1-level and 0-level

Rise Time

The duration of rising time from 20% to 80%

Fall Time

The duration of falling time from 80% to 20%

Eye Crossing

The measure of the amplitude of the crossing point relative to the one andzero level

Crossing(%)=

Signal-to-Noise

V1 − V0

The ratio of the desired signal level to the level of background noise.

Ratio

Q=

Average Power

VCrossin g − V0

V1 − V01 + 0

The average value of the whole data stream. A higher percentage of 1level comes with a higher average power. The figure comes to 50% of eyeamplitude when the 1-level and 0-level account for the same probability

262

int.siglent.com

SDS6000L User Manual

28.5

Mask Test

Click the Mask Test under the eye diagram menu to start the mask test configuration. Its principle
and operation process are the same as the mask test in normal mode. Refer to “Mask Test” to see thedetails.

28.6

Other Operation

Some operations are supported under both the eye diagram analysis and normal mode. Part of thecompatible operations are listed in the following table. For details of their operations, please refer totheir respective sections.

Table 28.5 Operations that be supported under the eye diagram
Operation

Note

Vertical Control

The vertical scale (volts/div) and vertical position can be
adjusted.

Horizontal

The horizontal scale (time/div) and trigger delay can beControl andadjusted. Memory depth will adjust according to the time

Samplingbase under the highest sampling rate.

Trigger Control

The trigger setting is supported. Keeping the default edge
trigger mode and trigger level to ensure the oscilloscope cantrigger correctly will be recommended.

Display

The color temperature, waveform intensity, and grid aresupported to adjust.

Cursors

Both vertical and horizontal cursors can be added andadjusted.

Systemint.siglent.com

System-related settings, I/O settings, etc.

263

SDS6000L User Manual

29 Jitter Analysis
29.1

Overview

Jitter — The time deviation from the ideal timing of a data-bit event — is one of the most important
topics in high-speed digital data signals. The Jitter analysis application featured in this device canmeasure kinds of jitter features. Besides, the various jitter components can be decomposed based on
TIE data and analyzed in time and frequency domains.

Click

Analysis>

Jitterto recall the jitter analysis dialog box, this process may cost several

seconds. Its user interface is similar to the UI in an eye diagram or normal mode.

A.

Menu BarB.

The grid area displays the waveform in normal or eye diagram modes.

C. Dialog Box
D. Measurement parameter display area

264

E.

Clock or data rate information areaF.

Timebase and Trigger descriptor box

int.siglent.com

SDS6000L User Manual

Jitter Analysis Dialog Box
A.

Turn on/off the Jitter Analysis

B.

Set the display mode, including Waveformand Eye diagram

C. The signal setting, including source andtrigger level
D. Quick view. After setting the channel
source and performing a quick view, theoscilloscope will automatically adjust thevertical/horizontal scale and trigger level,
the frequently-used measurement types
will be recalled as wellE.

Clock recovery options, including ConstantFrequency and PLL

F.

Jitter decomposition setting

G. Set the measurement

29.2

Signal Configuration

Click Signal to set the channel source and trigger level. The setting is similar to Eye Diagram. Referto the “Signal Setting” section in the "Eye Diagram" chapter for details.

The type of measured signal can be pre-set to be data or clock in the jitter analysis, then theoscilloscope will execute corresponding optimization processing for the signal type, which is different
from Eye Diagram.

int.siglent.com

265

SDS6000L User Manual

29.3

Clock Recovery

The setting of clock recovery is the same as Eye Diagram. Refer to the section "Clock Recovery" inthe "Eye Diagram" chapter.

29.4

Jitter Decomposition

Jitter typically consists of various jitter components that are caused by different jitter sources. For ananalytical approach, a jitter model is frequently used, which splits jitter into two major categories ofrandom and deterministic jitter.

Figure 29.1 Jitter decomposition tree

The benefits of jitter decomposition：
1.

Jitter performance plays a crucial role in the bit error rate of a high-speed digitalcommunication system. Through a limited sample, the jitter measurement value can be used
to estimate the TJ at any bit error rate if the RJ estimation is correct.

2.

Jitter decomposition is a key tool to accurately derive each type of jitter as well as total jitterin a system and identify the root causes of jitter quickly.

266

int.siglent.com

SDS6000L User Manual

Click Jitter Decomposition to enter the jitter decomposition configuration menu.
A.

Set the pattern mode between Auto andManual. In the Manual mode, the patternlength in bits, which represents the repetition
period of the data sequence, can be set as
user input. In the Auto mode, the patternlength is automatically detected

B.

Pattern length. When Pattern Mode = Auto,this area is read-only. The pattern length is anecessary parameter to calculate DCD and
DDJ

C. Enable/Disable the Bathtub plot. The "bottom"of the bathtub expresses the degree of theeye opening under the target bit error rate.

The result of jitter decomposition will be shown in the measured area. Users can add the jitter
components interested, including measurement values, histogram, track, and spectrogram.

29.5

Jitter Measure

Click the Jitter Measure in the Jitter analysis menu to recall the jitter measurement settings. The
operation method of adding jitter measurement parameters is the same as that of the measure closed
jitter function. Refer to the “Measurement” for details.

Statistics for each measurement item are automatically enabled when the measurement is activated.
Following are the descriptions of some jitter parameters.

int.siglent.com

267

SDS6000L User Manual

Table 29.1 Descriptions of jitter parameters
Parameter

Description

Jitter Basics

PeriodFrequency+Width-Width

+Duty

-Duty

CCJ

+Width@C2C-Width@C2C+Duty@C2C-Duty@C2C

The time between the crossing points of two consecutive like-polarity
edges. For clock data only.
Reciprocal of the period. For clock data only.
Time difference between the middle of a rising edge to the middle ofthe next falling edge of the pulse.
Time difference between the middle of a falling edge to the middle ofthe next rising edge of the pulse.
Positive Duty Cycle. The ratio of positive width to period. For clock
data only.
Negative Duty Cycle. The ratio of negative width to period. For clock
data only.
The difference between two continuous periods. For clock data only.
Difference between the positive pulse width of two continuous
cycles. For clock data only.
Difference between the negative pulse width of two continuous
cycles. For clock data only.
Difference between the positive duty cycle of two adjacent cycles.
For clock data only.
Difference between the negative duty cycle of two adjacent cycles.
For clock data only.

Decomposition

TIE

268

Time Interval Error (TIE) is the time difference between the signal
edge and the reference clock.

int.siglent.com

SDS6000L User Manual

Random jitter (RJ) also called Gaussian jitter, is unpredictableelectronic timing noise. RJ is caused by thermal noise in an electricalRJ

circuit or due to the central limit theorem. Random jitter is unbounded
and commonly specified by the standard deviation σ. Due to its
irregular nature, random jitter (RJ) is uncorrelated to any other signal
and unpredictable in timing behavior.

DJ

Deterministic jitter, the observed peak-to-peak value will not growover time, has a bounded distribution.
Duty cycle distortion (DCD) is the difference between mean TIE for

DCD

rising edges and mean TIE for falling edges. Asymmetrical rise-time
vs. fall-time and non-optimal choice of decision threshold will cause
DCD.
Data-dependent jitter (DDJ) has several discrete line distributions,

DDJ

mainly caused by ISI, the physical/electrical effect. There will bedifferent zero-crossing positions, due to transmission line effects likereflections and the step response of the system.
Periodic jitter (PJ) is mainly caused by periodic signal interference on

PJ

the board, including power supply ripple, clock crosstalk, etc.
Periodic interference can directly cause phase modulation of theobserved signal.
Total jitter (TJ) is computed with a required bit error rate (BER) for
the system:
TJ = DJ + 2Q𝐵𝐸𝑅 ∗ σRJ

TJ@BER

The value of Q𝐵𝐸𝑅 changes according to different bit error rates. A
common BER used in communication standards such as Ethernet is10−12, Q𝐵𝐸𝑅 = 7.05

Part of the measurements can be displayed in frequency domains. Click the Spectrum in the dialog
of jitter measure to enter the frequency domain setting.
The vertical and horizontal parameters can be adjusted for better observation.

int.siglent.com

269

SDS6000L User Manual

29.6

Other Operation

Some operations are supported under both the jitter analysis and normal mode. The compatible
operations are similar to the eye diagram, take Table 28.5 as a reference.

29.7

System Effect on Jitter Measure

The system's influence on the jitter measurement must be taken into consideration. It is important toevaluate the degree of error introduced in the jitter measurement according to the specifications of themeasurement system itself. The methods applied to the measuring process also play an important rolein the jitter measurement.

The system error formula of oscilloscope jitter measurement is as follows:2

 Noise 2
err =  + Jitter Sample Clock Slew rate Sample clock jitter and noise in the formula are possible from the oscilloscope itself. Besides, the slew
rate of the signal under test also is related to the system error according to the formula. The slew rateis inversely proportional to the error when the noise value does not change. This indicates that weshould use signals with edges as fast as possible during jitter testing, and keep the signal at full scaleof the oscilloscope by adjusting the vertical scale.

270

int.siglent.com

SDS6000L User Manual

30 Display
Display settings include display type of waveform, color, persistence, grid type, trace brightness,graticule brightness, etc.

Click the menu Display > Menu to recall the display dialog box.
A.

Switch the waveform display type to Vectors(line display) or DotsB.

Turn on or off the color grade

C. Set persistence
D. Clear display. The operation clears all
waveforms displaying on the screen andclears persistE.

Set menu auto-hide timeF.

Set the menu style to Embedded or Floating.
When set to Embedded, the grid area will becompressed horizontally to display thecomplete waveform when the dialog box isdisplayed. When set to Floating, it will
directly cover part of the grid area when the
dialog box is displayed

G. Select the grid type (Full Grid, Light Grid,and No Grid)
H. Set the trace intensity (0~100%)I.

Set the graticule brightness (0~100%)

J.

Customize the colors of waveforms

K.

Display the axis labels

int.siglent.com

271

SDS6000L User Manual

Display Type
There is no difference between the vector and dot display types when the number of samples of aframe is greater than 1250. When the sample number is below 1250, there are some differences.


Vectors: The samples are connected by lines (i.e. interpolated) and displayed. The interpolation
methods include linear interpolation and sin(x)/x interpolation. See the section of "Acquisition
Setup" for details of interpolation.



Dots: Displays the raw samples directly.

Vector Display

Dot Display

272

int.siglent.com

SDS6000L User Manual

Note: In the Run state, due to the high waveform update rate of theoscilloscope, the waveform displayed is the superposition of multiple
frames. Therefore, what is seen using the dots display is not the discrete
sampling points, but the display effect similar to equivalent sampling. Stop
the acquisition to view the original samples of each frame separately.

Dots display in Run state

Color Grade
Color grade uses color temperature to map the probability of the waveforms. The greater the probability
that the waveform appears in a pixel, the warmer the color of the pixel. The smaller probability, thecolder the color temperature of that pixel.

The picture below shows the change of color from cold to warm.

int.siglent.com

273

SDS6000L User Manual

Color grade On

Color grade Off

Set Persist
With persistence, the oscilloscope updates the display with new acquisitions but does not erase theresults of previous acquisitions in the specified period. All previous acquisitions are displayed with
reduced intensity. New acquisitions are shown in their normal color with normal intensity. In
combination with SDS6000L's high waveform update rate and persist function, in some cases
anomalies in the waveform can be found in a short time without complex trigger settings to improve
test efficiency. Below is an example to display glitches in a data sequence with infinite persistence.

When the display dialog box and persist are off, performing Display > Persistence can turn on
persist.
Click Persist in the display dialog box to set the persistence time.


Off: Turn off persist.

Variable persist time (100ms, 200ms, 500ms, 1s, 5s, 10s, 30s): Choose different persist
times. The oscilloscope updates with the newly acquired waveform display. Acquired
waveforms will be cleared after the corresponding time has expired.

274

int.siglent.com

SDS6000L User Manual

Infinite: Select “Infinite”, previous acquisitions will never be erased until a Clear Display
operation is performed.

Set Grid

Full Grid: Display 8*10 gridLight Grid: Display 2*2 grid

No Grid: Display without grid

Full Grid

Light Grid

int.siglent.com

275

SDS6000L User Manual

No Grid

Color Setting
The color setting supports user-defined trace colors. Click the color rectangle box to set in the pop-up
palette page. Click the Default area to restore the default color for all channels.

276

int.siglent.com

SDS6000L User Manual

Select a color directly on the palette and preview its hue, saturation, light, and RGB values or directly
modify the color parameters to customize the color. After the change, click OK to confirm.

Axis Label Setting
Turn on the axis label, the horizontal axis label will be displayed at the bottom of the grid, and thevertical axis label will be displayed at the left of the grid. There are two display modes of the axis labels:
moving and fixed.
Moving mode: When moving the waveform, the position of the axes moves with the waveform,
while the coordinates remain fixed.



Fixed mode: The position of the axes remains fixed, while the coordinates update as the
waveform is moving.

int.siglent.com

277

SDS6000L User ManualMoving mode

Fixed mode278

int.siglent.com

SDS6000L User Manual

31 Waveform Generator
31.1

Overview

The SDS6000L supports waveform/function generation by incorporating a built-in Waveform Generator
accessory combined with the software activation using option SDS6000L-FG.

The AWG functions include:

6 basic waveforms: Sine, Square, Ramp, Pulse, Noise, and DC

45 built-in arbitrary waveforms



Output frequency up to 25 MHz

-3 V ~ +3 V output amplitude range

Refer to the datasheet for the detailed specifications of the AWG.

SDS6000L-FG Option
For the installation of the SDS6000L-FG option, please refer to the section "Install Option".
Click the menu Utility > AWG Menu to recall the AWG dialog box.
A.

Turn on/off the output of AWG

B.

Select the waveform type (Sine, Square,
Ramp, Pulse, Noise, DC, and Arb)

C. Set the frequency
D. Set the amplitudeE.

Set the offset

F.

Other AWG settings: Output load,
overvoltage protection, and so on.

G. AWG system information and firmware
upgradeint.siglent.com

279

SDS6000L User Manual

31.2

Wave TypeThe AWG function provides six standard waveforms and multiple arbitrary waveforms. The standard
waveforms are Sine, Square, Ramp, Pulse, Noise, and DC.

The following table shows all waveform types and corresponding parameters.

Wave Type

Parameters

Sine

Frequency, Amplitude, Offset

SquareFrequency, Amplitude, Offset, Duty

Ramp

Frequency, Amplitude, Offset, Symmetry

Pulse

Frequency, Amplitude, Offset, Duty

DC

OffsetNoise

Stdev, MeanArb

Frequency, Amplitude, Offset, Arb TypeThe Arbitrary waveforms consist of two types: built-in waveforms and stored waveforms. Click
Arb Type in the AWG dialog box, and select the arbitrary in the pop-up window:

280

int.siglent.com

SDS6000L User Manual

There are 6 tabs in the window. Under each tab, a catalog of waveforms is listed. Built-in waveforms
are stored in Common, Math, Engine, Window, and Trigo. Stored waveforms are located in the Storedmenu.
Users can edit arbitrary waveforms using SIGLENT EasyWaveX PC software, send the stored
waveforms to the instrument through the remote interface, or import the stored waveforms through a
U disk.

31.3

Other Setting

Click Setting in the AWG dialog box torecall the "Other Setting" dialog box:
A.

Select the output load

B.

Turn on/off the OVP (Over Voltage
Protection)

C. Do zero adjust
D. Set to default
E.

Return to the previous menu

Output Load
The selected output load value must match the load impedance. Otherwise, the amplitude and offsetof the output waveform of AWG will be incorrect.

OVP
When OVP is enabled, the output will be turned off automatically once the protection condition is met.
The protection condition is when the absolute value on the output port is higher than 4 V ± 0.5 V. At
the same time, a warning message is displayed.

int.siglent.com

281

SDS6000L User Manual

Zero Adjust
The output zero error of the AWG can be calibrated by using auto mode or manual mode. The objectof the Zero Adjust is: when AWG outputs a 0 V DC signal, the measured mean value of thecorresponding channel is within ± 1mV.



Auto Steps: In this mode, the adjustment can only be performed with C1 of the oscilloscope.
Connect the output of AWG to C1 of the oscilloscope, click Auto , and the AWG will start toautomatically adjust. The scope will display “Zero adjust completed!” when the adjustment is done.



Manual Steps: The AWG can be manually calibrated through any channel on the oscilloscope.
Take C2 as an example:
1.

Connect the output of AWG to C2, open C2, set it to DC coupling, turn on the bandwidth limit,
and set the probe attenuation to 1X.

2.

Set the vertical scale of C2 to a small scale such as 1 mV/div. Turn on the measure and set
the parameter to the Mean of C2.

3.

Click Manual and roll the mouse wheel to adjust the compensation value, until the meanvalue of C2 is within ±1 mV, and then click Save .

31.4

282

Systemint.siglent.com

SDS6000L User ManualDevice Info – Includes Model, Serial Number, Firmware Version, and Hardware Version of the AWG
module.

UpgradeThe firmware here refers to the firmware of the built-in AWG module. The SDS6000L supports firmware
and configuration file upgrades for the AWG via a U disk. Follow the steps below:
1.

Copy the upgrade file (*.ADS) to the U disk.

2.

Insert the U disk into one of the USB host ports of the oscilloscope.

3.

Click Browse… to select the path of the upgrade file. See the chapter "Save/Recall" forthe detailed operation.

4.

Click

Upgradeto start the upgrade progress. A progress bar shows the percentage

finished.
5.

After the upgrade, reboot the oscilloscope.

6.

Enter the System dialog box again to check if the upgraded hardware version number isconsistent with the target version.

Warning: Don't cut off the power during the upgrade !

int.siglent.com

283

SDS6000L User Manual

32 Save/Recall
The SDS6000L supports saving setups, reference waveforms, screenshots, and waveform data files
to internal storage, external USB storage devices (e.g. U disk), or network storage. Saved setups andreference waveforms can be recalled as needed.

For details on network storage, refer to the section “Network Mapping”.

32.1 Save TypeThe SDS6000L supports save types: Setup, Reference, Image (*.bmp/*.jpg/*.png), Waveform Data(binary/CSV/MATLAB), and FileConverter Tool. It also supports saving the current setup as the default
setting. Here are brief descriptions of save types:

Setup
The default save type of the oscilloscope. The setup is saved with the *.xml file extension.

Reference
The reference waveform data are saved with the *.ref file extension. The saved file contains thereference waveform data and its setup information such as the vertical scale, vertical position, andtimebase.

BMP
Saves the screenshot in *.bmp format.

JPG
Saves the screenshot in *.jpg format.

PNG
Saves the screenshot in *.png format.

284

int.siglent.com

SDS6000L User Manual

Binary Data
Saves the waveform data in binary (*.bin) format.

CSV Data
Saves the waveform data in ".csv" format. After selecting this type, you can click the ParaSwitch todetermine whether to include the scope configuration parameter information (horizontal timebase,vertical scale, etc.) or not. Click the Save All Channel to save all displayed waveforms on the screen
or select the Source to save. The available sources include C1 ~ C8 and F1 ~ F4.

Matlab Data
Saves the waveform data in *.mat or *.dat format which can be imported by Matlab directly. The
available sources include C1 ~ C8 and F1 ~ F4 (except for FFT).

To Default Key
The oscilloscope provides two options for the default setting. When the Default Type is set to “Factory”,
the factory default setting will be recalled by performing Acquire > Default ; When the Default
Type is set to “Current”, the last setting saved by the “To Default Key” operation will be recalled.

FileConverter
The mini tool is used to convert stored binary files to CSV format for viewing with a spreadsheet
program. This is ideal when collecting large datasets. For a waveform frame with deep memory suchas 250 Mpts, saving directly as a CSV file will take a long time and will occupy a large amount ofmemory on a USB storage device. It’s recommended to save the data as a binary file and then convert
it to a CSV file on a computer.

To Memory
This is used to save the specified source to a memory trace (M1~M4). Refer to the chapter "Memory"for details.

int.siglent.com

285

SDS6000L User Manual

The following table shows the relationship between the save types and save/recall operations.

286

Type

Save to External

Recall

Setup

√

√

Reference

√

√

BMP

√

×

JPG

√

×

PNG

√

×

Binary Data√×

CSV Data√

×

Matlab Data√

×

FileConverter√×

Memory√×

int.siglent.com

SDS6000L User Manual

32.2

File Manager

The SDS6000L's file manager has a similar style and operation to the Windows© operating systems.

A.

Address bar

B.

Navigation panel

C. File list
D. Toolbar
E.

File typeF.

Close the file manager

Table 32.1 Description of icons in the file manager
Icon

int.siglent.com

Description

Icon

Description

Back

Forward

Up level

Root directory

Save

Save as287

SDS6000L User Manual

RecallNew directory

New file

Copy

Paste

RenameDelete32.3

Save and Recall Instances

Save the screenshot to the path “Udisk0\sds6l\” in the format of BMP
First, Insert the U disk.
Secondly, set the parameters of the Save operation:

Select Mode as "Save"

Select Type as "BMP"

Select the Image Style. "Normal" saves
images with the same color of the display;"Inverted" saves images with a whitebackground color to save ink during printingSelect to include the menu in the Print Areaor not. "No" only saves the grid area anddescriptor boxes; "Yes" saves the wholedisplay

Click File Manager to open the file manager

Third, select the “\Udisk0\sds6l\” directory in the file manager;
Fourth, click Save As , click the pop-up text box to recall the virtual keyboard and input the file
name, and then click OK :288

int.siglent.com

SDS6000L User ManualOnce saved, the new BMP file is visible in the file manager:

In step 4, if Save is selected instead of Save As , the system saves the file as the default filename SDS6xxxL_BMP_n.bmp, where n is an integer incrementing from 1. The default save path is\SIGLENT\.

Note: When a file is saved successfully, a message will pop up to inform thatit has been saved, and the filename of the next file to be saved is shown as
well. The default naming strategy is to add an incremental number at the endof the first saved filename.

int.siglent.com

289

SDS6000L User Manual

Note: A saved picture can be browsed on the oscilloscope by opening it in theFile Manager.

Recall the setup file " Track.xml" stored in path “Udisk0\sds6l\”
First, insert the U disk with the file " Track.xml" stored folder “\sds6l\”.
Second, set the parameters of the Recall operation:290

Select Mode as "Recall"Select Type as "Setup"

Click File Manager to open the file manager

int.siglent.com

SDS6000L User Manual

Third, select the “ \Udisk0\sds6l\ ” directory in the file manager, and then select the setup file
"Track.xml"

Fourth, click the Recall icon

int.siglent.com

and wait for the oscilloscope to finish recalling the setup.

291

SDS6000L User Manual

33 Utility
33.1

System Information

Operate Utility > Menu > System Info to check the system status. System information
includes the contents shown below.

33.2

System Setting

33.2.1

Language

The SDS6000L supports multiple languages, including Simplified Chinese, Traditional Chinese,
English, French, Japanese, German, Spanish, Russian, Italian, Portuguese, and so on.

Operate Utility > Menu > System Setting > Language , and select the language in the
list.

292

int.siglent.com

SDS6000L User Manual

33.2.2

Screen Saver

The screen saver will be activated if the oscilloscope has not been operated for a period. At this time
the output of the display is cut off to save power consumption.

Operate Utility > Menu > System Setting > Screen Saver to specify the period before thescreen saver activates, or select "Off" to disable the screen saver.

Any action from the mouse, web control, or front panel can trigger to exit the screen saver.

33.2.3Sound

Follow the steps Utility > Menu > System Setting > Beeper to enable or disable theaudible buzzer.

33.2.4

Auto Power-on

Operate Utility > Menu > System Setting > Auto Power On to set. See the chapter“Mechanical Dimension” for details.

33.2.5Date/Time

The SDS6000L has an RTC clock, which helps to record absolute time information for screenshots,history frames, and so on.

Operate Utility > Menu > System Setting > Date/Time to open the Date/Time dialog box:

int.siglent.com

293

SDS6000L User Manual

Click the text box of Hour, Minute, Second, Day, Month, and Year to edit these fields. Click the
Modify Date/ Time button to perform the change. Click the text box of Time Zone to select the timezone. Click the Modify Time Zone button to perform the change.

33.2.6

Reference Position Setting

The reference position is used to set the strategy of scaling on the horizontal and vertical axes,
adapting to different requirements.

Operate Utility > Menu > System Setting > Reference Pos to open the dialog box, andchoose the strategy for the offset value change in the vertical (or horizontal) direction when thevertical (or horizontal) scale is changed.

Horizontal Ref


Fixed Delay: When the time base is changed, the horizontal delay value remains fixed. As
the horizontal timebase scale is changed, the waveform expands/contracts around thecenter of the display



Fixed Position: When the time base is changed, the horizontal delay remains fixed to thegrid position on the display. As the horizontal time base scale is changed, the waveform
expands/contracts around the position of the horizontal display.

294

int.siglent.com

SDS6000L User Manual

Vertical Ref


Fixed Offset: When the vertical scale is changed, the vertical offset remains fixed. As thevertical scale is changed, the waveform expands/contracts around the main X-axis of thedisplay

Fixed Position: When the vertical scale is changed, the vertical offset remains fixed to thegrid position on the display. As the vertical scale is changed, the waveform expands/contracts
around the position of the vertical ground position on the display.

Take Horizontal Ref as an example to demonstrate the scaling effect of different settings:

Timebase = 10 ns/div, Horizontal Delay = -20 ns = - 2 div

int.siglent.com

295

SDS6000L User ManualFixed position, timebase is changed to 5 ns/div, the grid number of delay (-2div) remains
fixed, while the horizontal delay changes to -10 ns

Fixed delay, timebase is changed to 5 ns/div, the horizontal delay value remains fixed,
while the grid number of delay changes to -4div

296

int.siglent.com

SDS6000L User Manual

33.2.7

Tips

When a file is saved successfully, a message will pop up to inform the user that the image has beensaved. The Message box status can be set as shown below:

Operate Utility > Menu > System Setting > Tips to open the tips dialog box:


Hold: The message box will remain at the
top of the waveform display area.



Fade: The message box will disappear
automatically after the set time, and thetime range is: 0.1~9.9s.

None: The message box will not pop up.

33.3

I/O Setting

33.3.1LAN

Operate the following steps to set the LAN port:

Perform

Utility

int.siglent.com> I/O > LAN Config to open the LAN Config dialog box.

297

SDS6000L User Manual

A.

Check Automatic to enable dynamic IP. In
this case, the oscilloscope must be connectedto the local area network with a DHCP server.
Contact your network administrator to confirm
the relevant information.

B.

When

is not checked, the oscilloscope

uses static IP. Set the static IP address,
subnet mask, and gateway separately.
C. If your network provides Dynamic DNS, you
can enable Automatic (DNS) to let theoscilloscope register its hostname and use the
DNS server for name resolution.
D. Whenis not checked, set the DNS IP

address manually.
E.

When accessing more than two SIGLENT
instruments through the web browser, it isnecessary to set a different VNC port number
for each instrument. The range is from 5900 to
5999.

33.3.2Clock Source

The clock source of the device can be set to "Internal" or "External". When the clock source is set to"external", the device receives the 10 MHz clock from the "10 MHz In" port on the rear panel and uses
it as a reference to generate the sampling clock of the system. Both “internal" and "external" clock
sources can be output from the 10 MHz Out port for synchronizing other instruments.

When the clock source is set to "External", an icon indicating an external clock will appear in the lower
right corner of the display:
The clock source is external.
The clock source is external, but no valid external clock is detected.

298

int.siglent.com

SDS6000L User Manual

33.4

Install Options

The SDS6000L provides a few options to enhance its functionality. Contact your local SIGLENT sales
representative or SIGLENT technical support to get the corresponding option key. Perform thefollowing steps to install the option:
Utility > Menu > Software Options

A.

Option information display area. When the option is not activated, the license type is displayed as"Temporary" and can be tried up to 30 times.

B.

Select the option to install.

C. The option key input area, click the text box, and enter the key by the virtual keyboard.
D. Clear the characters in the key input area.
E.

After entering the option key, click Install to perform the installation.

F.

Use the U-disk to automatically install the option, the license must be stored in the root directoryof the U-disk.

int.siglent.com

299

SDS6000L User Manual

33.5

Maintenance33.5.1

UpgradeThe firmware is upgradeable through an external USB memory device/U disk. Make sure the U disk
contains the correct upgrade file (*.ads) and is connected to the oscilloscope before performing the
upgrade.

Operate Utility > Menu > Maintenance > Upgrade to recall the upgrade dialog box:

Click Browse to open the file manager, select the correct upgrade file and click the Recall icon

300.

int.siglent.com

SDS6000L User Manual

Click Upgrade in the upgrade dialog box to start the upgrade. The oscilloscope first copies the
upgrade file (*.ads) to the local memory and parses it. If the parse succeeds, it will show the following
dialog. Users can choose Cancel to cancel the upgrade or Reboot to restart the oscilloscope
immediately and continue the upgrade. Otherwise, the oscilloscope will restart automatically to finish
the upgrade.

After the reboot, check if the version number is in the "Information System".

Warning: Do not turn off the power during upgrading. Otherwise, theoscilloscope may not boot up!

33.5.2

Self-Calibration

The self-calibration program can quickly calibrate the oscilloscope to reach the best working state andthe most precise measurement. It is recommended to perform a self-calibration if the change of ambient
temperature is more than 5℃.

Note: Make sure the oscilloscope has been warmed up or operated for morethan 30 minutes before the self-calibration.

int.siglent.com

301

SDS6000L User Manual

Please do the self-calibration as follows:
1.

Disconnect everything from all inputs.

2.

OperateUtility

>

Menu

>

Maintenance>Self Calibration, and the following

dialog box appears. Select Continue to start the self-cal program.

3.

The oscilloscope will not respond to any operation until the self-cal is finished. After the selfcal is completed, click the screen to exit.

33.5.3Developer Options

This function is used for internal development by SIGLENT.

33.6

Service33.6.1Web

The SDS6000L includes a web server function, you can access and control the oscilloscope using a
web browser. See the section "Web Browser" for details about remote control.

Perform

Utility> Menu>Service>Webto set the password to control the instrument.

The password length is limited to 20 bytes.

33.6.2

Network Mapping

Perform Utility > Menu > Service > Network Mapping to open the network mapping dialog
box:

302

int.siglent.com

SDS6000L User ManualA.

Network mapped drive letter. The default is I:

B.

Input the network storage directory in the text box. Check “Remember” toremember it.

C. Check“Anonymous”to access the network directory in guest mode (Username:Guest, no password). This needs the server to permit anonymous access.
D. Username and password. Check “Remember” to save them.

33.6.3E.

Check to automatically connect to the network directory at every power on.

F.

Connect or disconnect to the network directory manually.

Emulation

When the “Emulation” option is set to “Tektronix”, the SCPI command set of the instrument is changed
to be compatible with Tek. This setting can help the user switch from a Tek scope with minimumchanges to the existing code.

int.siglent.com

303

SDS6000L User Manual

33.6.4

LXI

This oscilloscope complies with LXI version 1.5, defined in LXI device specification 2016, which canbe used to quickly build automated test systems.

Click LXI to recall the LXI status box as below:

Click LAN Reset to restore the LAN defaults to your oscilloscope. It will enable automatic IP,
ICMP Ping responder, and Multicast DNS, and reset the webserver password which is set in
Utility > Menu > Service > Web .

After setting the LAN, the LXI welcome page can be loaded through the browser, refer to the section“LAN” and “Web Browser” for details. When starting instrument identification in the web server, the
LXI image in the upper right corner will flash continuously.

304

int.siglent.com

SDS6000L User Manual

33.6.5

Share File

After setting the shared file, you can access the internal files of the device through Windows. Perform
Utility > Menu > Service > Share File to open the share file dialog box:

A.

Set the working status of the share file.

B.

The share directory IP address (same as the device IP address)

C. Username. The default is ShareUser and cannot be changed.
D. Password. It can only be modified when sharing is stopped.
E.

Check to automatically start up the share file at every power on.

F.

Check to reset share file.

After running the service, enter the shared file of the access device under Windows:

int.siglent.com

305

SDS6000L User Manual

34 Troubleshooting
The commonly encountered failures and their solutions are listed below. When you encounter those
problems, please solve them using the corresponding steps. If the problem remains, please contact
SIGLENT as soon as possible.

1.

The screen is still dark (no display) after power on:
1)

Check whether the power is correctly connected.

2)

Check whether the power switch is on.

3)

Check whether the fuse is burned out. If the fuse needs to be changed, please contact
SIGLENT timely and return the instrument to the factory for replacement by the
maintenance personnel authorized by SIGLENT.

4)

Restart the instrument after finishing the above inspections.

5)

If it still does not work correctly, please contact SIGLENT.

Note: It takes up to 1 minute from power on to display an image on an
external monitor.

2.

The signal is sampled but no waveform of the signal is displayed:
1)

Check whether the probe is correctly connected to the signal connecting wire.

2)

Check whether the signal connecting wire is correctly connected to the BNC (namely
channel connector).

3.

3)

Check whether the probe is correctly connected to the item to be tested.

4)

Check whether there are signals generated from the item to be tested.

5)

Resample the signal.

The tested voltage amplitude is greater or lower than the actual value (Note that this problem
usually occurs when the probe is used):1)

Check whether the attenuation coefficient of the channel complies with the attenuation
ratio of the physical probe.

2)

306

Disconnect the oscilloscope from the external signal and do a self-cal.

int.siglent.com

SDS6000L User Manual

4.

There is a waveform display but not stable:
1)

Check the trigger signal source: check whether the source item at the trigger panel
complies with the signal channel used.

2)

Check whether it is a "false wave”: when the signal frequency is very large (more thanhalf of the sample rate), it is easy to appear as a "false wave". At this point, a small
timebase should be set to make the sample rate more than 2 times the signal frequency.

3)

Check the trigger type: general signals should use the “Edge” trigger and video signals
should use the “Video” trigger. Only when the proper trigger type is used, can the
waveform be displayed stably.

4)

5.

Change the trigger holdoff setting.

No display after performing Acquire > Run :Check whether the mode at the trigger panel (TRIGGER) is on “Normal” or “Single” and whether
the trigger level exceeds the waveform range. If yes, set the trigger level to the middle or set themode to “Auto”.

6.

The USB storage device cannot be recognized:
1)

Check whether the USB storage device can work normally.

2)

Make sure the USB interface can work normally.

3)

Make sure that the USB storage device being used is a flash storage type. Thisoscilloscope does not support hardware storage type.

4)

Make sure that the U disk system format is FAT32.

5)

Restart the instrument and then insert the USB storage device to check it.

6)

If the USB storage device still cannot be used normally, please contact SIGLENT.

int.siglent.com

307

About SIGLENTSIGLENT is an international high-tech company, concentrating on R&D, sales,
production and services of electronic test & measurement instruments.
SIGLENT first began developing digital oscilloscopes independently in 2002.
After more than a decade of continuous development, SIGLENT has extended
its product line to include digital
oscilloscopes,

function/arbitrary

oscilloscopes, isolated handheld

waveform

generators,

RF/MWsignalgenerators, spectrum analyzers, vector network analyzers, digital multimeters,
DC power supplies, electronic loads and other general purpose test
instrumentation. Since its first oscilloscope was launched in 2005, SIGLENT
has become the fastest growing manufacturer of digital oscilloscopes. We
firmly believe that today SIGLENT is the best value in electronic test &measurement.

Headquarters:
SIGLENT Technologies Co., Ltd
Add: Bldg No.4 & No.5, Antongda Industrial

Zone, 3rd Liuxian Road, Bao'an District,
Shenzhen, 518101, China
Tel: + 86 755 3688 7876
Fax: + 86 755 3359 1582
Email: sales@siglent.com
Website: int.siglent.com

North America:
SIGLENT Technologies America, Inc
6557 Cochran Rd Solon, Ohio 44139
Tel: 440-398-5800
Toll Free: 877-515-5551
Fax: 440-399-1211

Email: info@siglentna.com
Website: www.siglentna.com

Europe:SIGLENT Technologies Germany GmbH
Add: Staetzlinger Str. 70
86165 Augsburg, Germany
Tel: +49(0)-821-666 0 111 0
Fax: +49(0)-821-666 0 111 22
Email: info-eu@siglent.com
Website: www.siglenteu.com

Follow us on
Facebook: SiglentTech