SPECIFICATIONS for CHESTAC-8900(D-type)

SPECIFICATIONS
Product Type
Volume Detector
Volume Range
Flow Detection
Flow RangeFrequency
Accuracy
Gas Control
Gas Ejection
Gas Meter Calibration
Flow Sensor Calibration
PCPrinter
Data transmission
FRC test
Co Analyzer
He Analyzer
Power Supply
Power Consumption
Dimensions
Weight: Closed Circuit Pulmonary Function Test
: Rolling seal type spirometer: 0 ~ 10 L
: Volume differential method: 0 ~ +/- 14 L/s: 0 ~ 10 Hz: Within +/-2% or +/-50ml, whicheveris greater: Automated computer control: Automated washout: Automated calibration: Not required: Windows PC: Inkjet color printer: RS-232C: Helium dilution method: Electrochemical cell method
Range: 0 ~ 0.33%
Response: 30 sec.
: Thermal conductivity type catharometer
Range: 0 ~ 10%
Response: 10 sec.
: AC 220V or 110V, 50/60 Hz: Approx. 550VA(main unit): 770(W) x 690(D) x 800(H) mm: Approx. 110kg

STANDARD ACCESSORIES
Breathing tube
3-way cockRemote switch
Paper mouthpiece (100pcs/box)
Silicon mouthpiece
Spiro filter (Standard, Mouthpiece type)
Nose clip
CO2 Absorbents (1L)
Silica gel (500g): 4 pcs: 1 pc: 1 pc: 1 box: 4 pcs: 3 pcs each
: 2 pcs: 2 pcs: 1 pcCOMPREHENSIVEPULMONARY FUNCTION TEST LAB

ANALYZED ITEMS
VC, FVC, FV, MVV, MV, BD TEST, BMR,
FRC (Multiple Breath Helium Dilution)
DLCO (Single Breath Krogh Technique)

Printout sample

Spiro-Filter designed to fit on CHESTAC-8900
All specifications subject to change without notice
MANUFACTURED BY

3-25-11 Hongo Bunkyo-ku Tokyo, 113-0033 JAPAN
Telephone : (03) 5804-5031 Fax : (03) 3812-7220

DISTRIBUTED BY:

Highly Accurate and Reproducible
Volume type Spirometer

Reliability and Accomplishment
Since 1965…
Here comes

ALL NEW CHESTAC!
The CHESTAC-8900 is a fully automated pulmonary testing system for
Spirometry, FRC, and DLCO (Type D), + CV and N2WO (Type DN), featuring
reliable and time proven dry rolling seal spirometer, a volumetric spirometer
which offers you the highest level of measurement accuracy. It assists
operator with guiding messages in evaluating patient performance,
reproducibility of test result, and generating the report, which makes all test
procedures simpler and faster.

Use-friendly software
Printout format, analysis items, predicted formula and data-transmission
items can be selected and set to intended use. The software also includes
3 test result memory with waveform for comparison and BD testing. All test maneuvers andperformances are based upon the recommendations by American Thoracic Society.

Abundant Utility software

Auto Hardware Check

Statistical Analysis

Adopted the reliable rolling seal type spirometer
Our spirometer has received good reputation as a reliable mechanics for detectionof volume with enough experience and of first-class capability. The inside can be cleaned
and sterilized very easily. Ultraviolet-lamp and inner gas washout function are installed,which keeps the device very sanitary. Because of the fixed excursion volume displacement
ratio, there is no need for recalibration on site.
VOLUMETRIC SPIROMETER
requires NO CALIBRATION

Easy Cleanliness and Sterilization
to Reduce the risk of Patient Cross Contamination

Washable Test Balloons

FVC Test Screen

Accuracy Management

Trend Graph
Detachable 3 way cock for sterilization

ComputerDiagnostic Screen

Incentive Screenfor Children
UV lamp for Sterilization, equipped in the Spirometer Bell