0101010101010101010101010101010101010101010101
0101010101010101010101010101010101010101010101
0101010101010101010101010101010101010101010101
0101010101010101010101010101010101010101010101
0101010101010101010101010101010101010101010101
0101010101010101010101010101010101010101010101
0101010101010101010101010101010101010101010101
0101010101010101010101010101010101010101010101
0101010101010101010101010101010101010101010101
0101010101010101010101010101010101010101010101
Digital Oscilloscope
0101010101010101010101010101010101010101010101
Handheld Oscilloscope
0101010101010101010101010101010101010101010101
0101010101010101010101010101010101010101010101
Waveform Generator
0101010101010101010101010101010101010101010101
DC Power Supply
0101010101010101010101010101010101010101010101
DC Electronic Load
0101010101010101010101010101010101010101010101
Digital Multimeter
0101010101010101010101010101010101010101010101

Probes & Accessories

SIGLENT TECHNOLOGIES CO., LTD

SIGLENT TECHNOLOGIES产品手册PRODUCT CATALOG

Catalog

CatalogCompany Profile

2

SDS7000A Super Phosphor Oscilloscope

3

SDS6000A Super Phosphor Oscilloscope

10

SDS6000L Low Profile Oscilloscope

15

SDS5000X Super Phosphor Oscilloscope20

SDS3000X HD Super Phosphor Oscilloscope24

SDS2000X HD Super Phosphor Oscilloscope29

SDS2000X Plus Super Phosphor Oscilloscope

35

SDS2000X-E Super Phosphor Oscilloscope

39

SDS1000X HD Super Phosphor Oscilloscope

43

SDS800X HD Super Phosphor Oscilloscope

47

SDS1000X-E Super Phosphor Oscilloscope

52

SDS1000DL+/CML+ Digital Storage Oscilloscope

56

SHS1000X/SHS800X Handheld Oscilloscope

58

SDG7000A Arbitrary Waveform Generator

64

SDG6000X Pulse/Arbitrary Waveform Generator

71

SDG2000X Function/Arbitrary Waveform Generator

78

SDG1000X Function/Arbitrary Waveform Generator

83

SDG800 Function/Arbitrary Waveform Generator

87

SPS5000X Programmable Switching DC Power Supply

89

SPD3000 Programmable Linear DC Power Supply

93

SPD1000X Programmable Linear DC Power Supply

95

SDL1000X DC Electronic Load98

SDM3065X Digital Multimeter102

SDM3055 Digital Multimeter105

SDM3045X Digital Multimeter

108

Probes & Accessories

111

Other Products Overview

116

Service

119* Spetrum Analyzer, Network Analyzer, RF Signal Generator can be found in SIGLENT RF PRODUCTSCATALOG

modification date：2024-03
www.siglent.com 1

Company Profile

SIGLENT TECHNOLOGIES Co., Ltd.
Every Bench. Every Engineer. Every Day.
SIGLENT has been providing test & measurement solutions for almost 18 years from its headquarterin Shenzhen, China. There are more than 300 employees, one third of whom are high-educated R&D
engineers.
SIGLENT has many patent technologies. We are dedicated to develop sophisticated and high quality
digital oscilloscopes, waveform generators, RF signal generators, handheld digital oscilloscopes, spectrum
analyzers, vector network analyzers and DC power supplies, DC Electronic Loads, digital multimeters. We
strive to deliver the highest quality of customer service and satisfaction to our customers.

SIGLENT provides the following
instruments:-Digital Oscilloscope-Handheld Oscilloscope-Waveform Generator-RF Signal Generator-Spectrum Analyzer-Vector Network Analyzer-DC Power Supply-DC Electronic Load
-Digital Multimter-Probes & Accessories

SIGLENT sincerely invite
you to join
Please email :sales@siglent.com

www.siglent.com 2

Digital Oscilloscope

SDS7000A
Super Phosphor Oscilloscope
Key Features• 4 analog channels, up to 4 GHz bandwidth with up to 20 GSa/s sample rate
• 12-bit ADC• Low background noise: 220 μVrms @ 4 GHz bandwidth• SPO technology• Waveform capture rates up to 1,000,000 wfm/s• Supports 256-level intensity grading and color temperature display modes• 500 Mpts/ch standard, 1 Gpts/ch optional• Digital trigger system• Intelligent trigger: Edge, Slope, Pulse, Window, Runt, Interval, Dropout, Pattern, Qualified, Nth edge, Setup/hold, Delay and Video (HDTV
supported). Zone Trigger simplifies advanced triggering• Serial bus triggering and decoder, supports protocols I2C, SPI, UART, CAN, LIN, CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT, Manchester,
ARINC429 and USB 2.0• Segmented acquisition (Sequence) mode, dividing the maximum record length into multiple segments (up to 124,000), according to trigger
conditions set by the user, with a very small dead time between segments to capture the qualifying event• History waveform record (History) function, the maximum recorded waveform length is 124,000 frames• Automatic measurements on 50+ parameters, supports statistics with histogram, track, trend, Gating measurement, and measurements
on Math, History and Memory traces• 4 Math traces (32 Mpts FFT, Filter, addition, subtraction, multiplication, division, integration, differential, square root, etc.), supports formula
editor• Abundant data analysis functions such as Search, Navigate, SignalScan, Digital Voltmeter, Counter, Waveform Histogram, Bode plot, Power
Analysis, Eye/Jitter Analysis and Compliance Test
• High Speed hardware-based Average, Hi-Res; High Speed hardware-based Mask Test function, with Mask Editor tool for creating userdefined masks• 16 digital channels (optional)• Built-in 50 MHz waveform generator• Large 15.6” HD TFT-LCD display with 1920 * 1080 resolution; Capacitive touch screen supports multi-touch gestures• Interfaces include: 4x USB Host 3.1 Gen 1, 2x USB 3.0 Host, USB 2.0 Device, 2x 1000M LAN, DVI-D, DP 1.2, HDMI 1.4, Audio, External
Triger In, Aux Out (Pass/Fail, Trigger Out), 10 MHz In, 10 MHz Out• Built-in web server supports remote control over the LAN port using a web browser. Supports SCPI remote control commands. Supports
external mouse and keyboard

www.siglent.com 3

Digital Oscilloscope
Characteristics• 12-bit High Resolution

12-bit resolution shows you more details and less noise on the waveform.

Upgraded processor system
Processor fully upgraded from the embedded ARM processor to
the X86 processor, has greatly improved the system response
speed and the speed of measurement, calculation, and analysis,presenting more possibilities for the expansion of software
analysis functions in the future.

Excellent User Interface and User Experience• 15.6” HD display with 1920*1080 resolution• Capacitive touch screen, supporting multi-touch gestures,can move or scale the waveform traces quickly by fingertouch movements, which greatly improves the operational
efficiency• Built-in WebServer supports remote control on a web page
over LAN• Supports external mouse and keyboard

www.siglent.com 4

Digital Oscilloscope• High Waveform Update Rate

• Deep Record LengthWith a waveform update rate of up to 1,000,000 wfm/s, the oscilloscope

Using hardware-based Zoom technique and record length of up to 1

can easily capture unusual or low-probability events. In Sequence mode,Gpts, users can select a slower timebase without compromising thethe waveform capture rate can reach 1,100,000 wfm/s

sample rate, and then quickly zoom in to focus on the area of interest• Multiple Trigger Functions• Zone Trigger

Edge, Slope, Pulse, Video, Windows, Runt, Interval, Dropout, Pattern,

Zone Trigger is available for advanced triggering. Combine spatial

Qualified, Nth edge, Setup/hold, Delay and serial trigger

triggering with common trigger modes to isolate signals of interest• Advanced Math Function• Deep Memory FFT

In addition to the traditional (+, -, X, /) operations, FFT, Filter,

FFT supports up to 32 Mpts operation. This provides high-frequency

integration, differential, square root, and more are supported. Formularesolution with a fast refresh rate. The FFT function also supports a

Editor is available for more complex operations. 4 math traces arevariety of window functions so that it can adapt to different spectrum

available.

measurement needs. Three modes (Normal, Average, and Max hold) cansatisfy different requirements for observing the power spectrum. Autopeak detection and markers are supported.

www.siglent.com 5

Digital Oscilloscope• Measurements of a Variety of Parameters• Parameter Statistics Function

Parameter measurements include 4 categories: horizontal, vertical,

Statistics show the current value, maximum value, minimum

miscellaneous, and CH delay providing a total of 50+ different types

value, standard deviation, and mean value of up to 12 parametersof measurements. Measurements can be performed within a specified

simultaneously. A histogram is available to show the probability

gate period. Measurements on Math, Reference, and History frames are

distribution of a parameter. Trend and Track are available to show the

supportedparameter value vs. time.
For horizontal parameters such as period, all results are extracted from aframe, instead of just calculating the first one. This accelerates statistics
on horizontal measurements and enables distribution observation in aframe using Histogram and Track• History Mode

• Sequence Mode

History function can record up to 124,000 frames of waveforms. The

Segmented memory collection will store the waveform into multiple

recording is executed automatically so that the customer can playback

memory segments (up to 124,000) and each segment will store athe history waveforms at any time to observe unusual events and quickly

triggered waveform as well the dead time information. The interval

locate the area of interest using the cursors or measurements. The failed

between segments can be as small as 0.9 μs. All of the segments can beframes of the Mask Test can be stored as history

played back using the History function• Hardware-based High Speed Mask Test Function

The oscilloscope utilizes a hardware-based Mask Test function,
performing up to 80,000 Pass / Fail decisions each second. It is easy togenerate user-defined test templates to provide trace mask comparisons,making it suitable for long-term signal monitoring or automatedproduction line testingwww.siglent.com 6

Built-in Mask Editor application helps to create custom masks

Digital Oscilloscope

Compliance Test (Optional)
USB 2.0, 100Base-TX, 1000Base-T, 100Base-T1, 1000Base-T1 protocol
conformance testing are available. When the user sets up the environment
according to the prompts, by using the related test fixture, the oscilloscopeand related instruments can be automatically set up and related
measurement, calculation, decoding and other functions will be used fortesting, helping the user to complete each test project quickly and efficiently,
and reports are generated automatically.

• Bode Plot

• Power Analysis (Optional)

The oscilloscope can control the Built-in waveform generator, SIGLENTThe Power Analysis option provides a full suite of power measurements

isolated USB AWG module or a stand-alone SIGLENT SDG generator,

and analysis, which greatly improve the measurement efficiency into scan the amplitude and phase-frequency response of the DUT,

switching power supplies and power devices designand display the data as a Bode Plot. This makes it possible to replace
expensive network analyzers in some applications• Digital Channels/MSO (Optional)• Builit-in 50 MHz Function/Arbitrary Waveform
Generator (Optional)

Four analog channels plus 16 digital channels enable users to acquire

The oscilloscope can control the built-in waveform generator to output

and trigger the waveforms then analyze the pattern, simultaneously withwaveform with up to 50 MHz frequency and ±3 V amplitude. Six basic

one instrument

waveforms plus multiple types of arbitrary waveforms are built-in
www.siglent.com 7

Digital Oscilloscope

5 GHz Active Differential Probe
The SAP5000D differential probe is provided with 5GHz bandwidth, 80 ps
rise time, 400 fF differential input capacitance, and 10:1 attenuation ratio

Complete Connectivity• 2x USB 3.0 Host, 4x USB Host 3.1 Gen 1, USB 2.0 Device, USBTMC, 2x 1000M LAN (VXI-11+SCPI, Telnet (5024)+SCPI, Socket (5025) +SCPI, LXI, WebServer)• 1x DVI-D: up to 1920x1200 @ 60Hz, 1x DP 1.2: up to 4096x2304 @ 60Hz, 1x HDMI 1.4: up to 4096x2160 @ 60Hz
• Mic input, Audio Output• External Trigger In, Aux Out (TRIG OUT, PASS/FAIL), 10 MHz In, 10 MHz Out

Specifications
Model

SDS7404A H12

Analog channels

4 + EXT

Bandwidth

4 GHz

Sample rate (Max.)
Vertical Resolution
Memory depth (Max.)

SDS7304A H12
3 GHz

20 GSa/s (dual-channel)
10 GSa/s (3 or 4 channels)
12-bitUp to 16-bit in ERES modeStandard: 500 Mpts/ch
Optional: 1 Gpts/ch in dual-channel mode

Waveform capture rate (Max.) 1,000,000 wfm/s
Trigger type
Serial trigger and decode
Measurement

Edge, Slope, Pulse width, Window, Runt, Interval, Dropout, Pattern, Video, Qualified, Nth edge, Setup/hold, Delay, Serial
Standard: I2C, SPI, UART, CAN, LIN
Optional: CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT, Manchester (decode only), ARINC429, USB 2.0 (decode only)
50+ parameters, statistics, histogram, trend, and track supported
4 tracesMath

32 Mpts FFT, +, -, x, ÷, ∫dt, d/dt, √, Identity, Negation, Absolute, Sign, ex, 10x, ln, lg, Interpolation, MaxHold, MinHold, ERES,
Average, Filter. Supports formula editor

Data analysis
Digital channel (optional)
Waveform generator(optional)
www.siglent.com 8

Search, Navigate, History, Mask Test, Digital Voltmeter, Counter, Waveform Histogram, Bode plot and Power Analysis, Eye/Jitter
Analysis, SignalScan, Compliance Test (USB 2.0, 100Base-TX, 1000Base-T, 100Base-T1, 1000Base-T1)
16-channel; maximum sample rate up to 1 GSa/s; record length up to 50 Mpts
Builit-in, frequency up to 50 MHz, 125 MSa/s sample rate, 16 kpts waveform memory

Digital Oscilloscope
Processor System

Intel Core i3-8100 or better, 32GB memory, 250GB storage, Linux operating system
I/O: 4x USB Host 3.1 Gen 1, 2x USB 3.0 Host, USB 2.0 Device (USBTMC), 2x 1000M LAN (VXI-11+SCPI, Telnet (5024)+SCPI,
Socket (5025) +SCPI, LXI, WebServer)

Data analysis

Display: 1x DVI-D: up to 1920x1200 @ 60Hz; 1x DP 1.2: up to 4096x2304 @ 60Hz; 1x HDMI 1.4: up to 4096x2160 @ 60Hz
Audio: Mic input, Audio Output
Others: External Trigger In, Aux Out (TRIG OUT, PASS/FAIL), 10 MHz In, 10 MHz Out

Probe (Standard)

500 MHz, 1 probe supplied for each channel

Display

15.6” HD TFT-LCD with capacitive touch screen (1920*1080)

Ordering Information
Model

Description

SDS7404A H12

4 GHz, 20 GSa/s, 4-CH, 12-bit, standard 500 Mpts/ch memory depth, 15.6” capacitive touch screen

SDS7304A H12

3 GHz, 20 GSa/s, 4-CH, 12-bit, standard 500 Mpts/ch memory depth, 15.6” capacitive touch screenStandard Accessories

Quantity

USB cable

1

Quick start

1

Passive probe (SP3150A)

1/channel

Certificate of calibration

1

Wireless mouse

1

Power cord

1

Protective Cover1

Optional Accessories

Part No.

16-channel logic probe

SPL2016

Power Analysis deskew fixture

DF2001A

USB 2.0 test fixture

FX-USB2

Ethernet test fixture

FX-ETH

Automotive Ethernet test fixture

FX-AMETH

STB3 demo signal source

STB3

USB-GPIB adapter

USB-GPIB

High-speed active probe

SAP1000, SAP2500

High voltage probe

HPB4010High-speed differential probe

SAP2500D, SAP5000D

High voltage differential probe

DPB1300/DPB4080/DPB5150/ DPB5150A/DPB5700/DPB5700A

Current probe

CPL5100/CP4020/CP4050/CP4070/CP4070A CP6030/CP6030A/CP6150/CP6500

Transit case

CASE-S2

Options

Part No.

Waveform generator (software)

SDS7000A-FG

16 digital channels (software)

SDS7000A-16LA

Power Analysis (software)

SDS7000A-PA

Eye Diagram/Jitter Analysis (software)

SDS7000A-EJ

I2S trigger & decode (software)

SDS7000A-I2S

MIL-STD-1553B trigger & decode (software)

SDS7000A-1553B

FlexRay trigger & decode (software)

SDS7000A-FlexRay

CAN FD trigger & decode (software)

SDS7000A-CANFDSENT trigger & decode (software)

SDS7000A-SENT

Manchester decode (software)

SDS7000A-Manch

ARINC429 trigger & decode (software)

SDS7000A-ARINC

USB 2.0 decode (software)

SDS7000A-USB2

USB 2.0 compliance test (software)

SDS7000A-CT-USB2

100Base-TX compliance test (software)

SDS7000A-CT-100BASE-T

1000Base-T compliance test (software)

SDS7000A-CT-1000BASE-T

100Base-T1 compliance test (software)

SDS7000A-CT-100BASE-T1

1000Base-T1 compliance test (software)

SDS7000A-CT-1000BASE-T1

1Gpts memory depth (software)

SDS7000A-1GPTS

OCXO timebase (Assembled and calibrated in factory only)

10M_OCXO_L
www.siglent.com 9

Digital Oscilloscope

SDS6000A
Super Phosphor Oscilloscope
Key Features• 4 analog channels, up to 2 GHz bandwidth with 5 GSa/s (10 GSa/s ESR) sample rate at each channel• Low background noise, supports 0.5 mV/div to 10 V/div vertical scales• SPO technology• Waveform capture rates up to 170,000 wfm/s (normal mode), and 750,000 wfm/s (sequence mode)• Supports 256-level intensity grading and color temperature display modes• 500 Mpts Record length in total for all 4 channels• Digital trigger system• Intelligent trigger: Edge, Slope, Pulse, Window, Runt, Interval, Dropout, Pattern, Qualified, Nth edge, Setup/hold, Delay and Video (HDTV
supported). Zone Trigger simplifies advanced triggering• Serial bus triggering and decoder, supports protocols I2C, SPI, UART, CAN, LIN, CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT and
Manchester• Segmented acquisition (Sequence) mode, dividing the maximum record length into multiple segments (up to 80,000), according to trigger
conditions set by the user, with a very small dead time between segments to capture the qualifying event• History waveform record (History) function, the maximum recorded waveform length is 80,000 frames• Automatic measurements on 50+ parameters, supports statistics with histogram, track, trend, Gating measurement, and measurements
on Math, History and Ref• 4 Math traces (8 Mpts FFT, addition, subtraction, multiplication, division, integration, differential, square root, etc.), supports formula editor• Abundant data analysis functions such as Search, Navigate, Digital Voltmeter, Counter, Waveform Histogram, Bode plot, Power Analysis andEye/Jitter Analysis• High Speed hardware-based Average, Hi-Res; High Speed hardware-based Mask Test function, with Mask Editor tool for creatinguser-defined masks• 16 digital channels (optional)• 25 MHz function / arbitrary waveform generator, built-in multiple predefined waveforms• Large 12.1” TFT-LCD display with 1280 * 800 resolution; Capacitive touch screen supports multi-touch gestures• Interfaces include: USB Hosts, USB Device (USBTMC), LAN(VXI-11/Telnet/Socket), micro SD card, Pass/Fail, Trigger Out, HDMI
• Built-in web server supports remote control over the LAN port using a web browser. Supports SCPI remote control commands. Supports
external mouse and keyboard
www.siglent.com 10

Digital Oscilloscope
Characteristics

Excellent User Interface and User Experience• 12.1” display with 1280*800 resolution• Capacitive touch screen, supporting multi-touch gestures,canmove or scale the waveform traces quickly by finger-touch
movements, which greatly improves the operation efficiency• Built-in WebServer supports remote control on a web page
over LAN• Supports external mouse and keyboard• High Waveform Update Rate

• Deep Record LengthWith a waveform update rate of up to 170,000 wfm/s, the oscilloscope

Using hardware-based Zoom technique and record length of up to 500

can easily capture unusual or low-probability events. In Sequence mode,Mpts, users can select a slower timebase without compromising thethe waveform capture rate can reach 750,000 wfm/s

sample rate, and then quickly zoom in to focus on the area of interest• Multiple Trigger Functions• Trigger Zone

Edge, Slope, Pulse, Video, Windows, Runt, Interval, Dropout, Pattern,

Trigger Zone is available for advanced triggering

Qualified, Nth edge, Setup/hold, Delay and serial trigger

www.siglent.com 11

Digital Oscilloscope• Hardware-based High Speed Mask Test Function

The oscilloscope utilizes a hardware-based Mask Test function,

Built-in Mask Editor application helps to create custom masks

performing up to 18,000 Pass / Fail decisions each second. It is easy togenerate user-defined test templates to provide trace mask comparisons,making it suitable for long-term signal monitoring or automatedproduction line testing• Eye/Jitter Analysis

Supports eye diagram and jitter analysis/measurement. It can automatically extract the embedded reference clock from serial data and create the eye diagram.
Measurement on multiple eye/jitter parameters is provided. Mask test on eye diagrams is supported• Bode Plot

• Power Analysis (Optional)

The oscilloscope can control the isolated USB AWG module or a stand-

The Power Analysis option provides a full suite of power measurements

alone SIGLENT SDG generator, to scan the amplitude and phase-and analysis, which greatly improve the measurement efficiency infrequency response of the DUT, and display the data as a Bode Plot.

switching power supplies and power devices design

This makes it possible to replace expensive network analyzers in some
applications
www.siglent.com 12

Digital Oscilloscope• Digital Channels / MSO (Optional)• Serial Bus Decode

Display the decoded characters through the events list. Bus protocol
information can be quickly and intuitively displayed in tabular form. I2C,

Four analog channels plus 16 digital channels enable users to acquire

SPI, UART, CAN, LIN, CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT, andand trigger the waveforms then analyze the pattern, simultaneously with

Manchester are supported

one instrument• Complete Connectivity

USB Host 3.0 x2, USB Host 2.0 x2, USB Device 2.0 (USBTMC) x1, LAN (VXI-11/Telnet/Socket) x1, micro SD card x1, Auxiliary output (Pass/Fail, Trigger Out)x1 andHDMI x1

Specifications
Model

SDS6204A

SDS6104A

SDS6054A

Bandwidth

2 GHz

1 GHz

500 MHz

Sampling rate (Max.)

5 GSa/s (10 GSa/s ESR) @ each channel

Analog channels

4 + EXT

Memory depth (Max.)

500 Mpts/ch(single-channel), 250 Mpts/ch (dual-channel) , 125 Mpts/ch (3 or 4 channels)

Waveform capture rate (Max.)

Normal mode: 170,000 wfm/s; Sequence mode: 750,000 wfm/s

Vertical resolution

8-bit, up to 16-bit in Hi-Res mode

Trigger type

Edge, Slope, Pulse width, Window, Runt, Interval, Dropout, Pattern, Video, Qualified, Nth edge, Setup/hold, Delay, Serial

Serial trigger and decode
Measurement

Standard: I2C, SPI, UART, CAN, LIN
Optional: CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT, Manchester (decode only)
50+ parameters, statistics, histogram, trend, and track supported
4 tracesMath

8 Mpts FFT, +, -, x, ÷, ∫dt, d/dt, √, Identity, Negation, Absolute, Sign, ex, 10x, ln, lg, Interpolation, MaxHold, MinHold.
Supports formula editor

Data analysis
Digital channel (optional)

Search, Navigate, History, Mask Test, Digital Voltmeter, Counter, Waveform Histogram, Bode plot and Power Analysis, Eye/
Jitter Analysis
16-channel; maximum sample rate up to 1 GSa/s; record length up to 50 Mpts
www.siglent.com 13

Digital OscilloscopeSingle-channel external USB isolated waveform generator, frequency up to 25 MHz, 125 MSa/s sample rate, 16 kpts

Waveform generator (optional)

waveform memory
USB 3.0 Host x2, USB 2.0 Host x2, USB 2.0 Device, LAN, micro SD card, HDMI, External trigger, Auxiliary output

I/O(TRIG OUT, PASS/FAIL)

Probe (Standard)

SP3150A, 500 MHz, 1 probe supplied for each channel

Display

12.1 TFT-LCD with capacitive touch screen (1280*800)

Ordering Information
Model

Description

SDS6204A

2 GHz, 10 GSa/s, 4-CH, 500 Mpts/ch memory depth, 12.1” capacitive touch screen

SDS6104A

1 GHz, 5 GSa/s, 4-CH, 500 Mpts/ch memory depth, 12.1” capacitive touch screen

SDS6054A

500 MHz, 5 GSa/s, 4-CH, 500 Mpts/ch memory depth, 12.1” capacitive touch screenStandard Accessories

Quantity

USB cable

1

Quick start

1

Passive probe (SP3150A)

1/channel

Certificate of calibration

1

Wireless mouse

1

Power cord

1

Optional Accessories

Part No.

Waveform generator (software)

SDS6000Pro-FG

25 MHz isolated USB function/arbitrary waveform generator

SAG1021I

16 digital channels (software)

SDS6000Pro-16LA

16-channel logic probe

SPL2016

Power Analysis (software)

SDS6000Pro-PA

Power Analysis deskew fixture

DF2001A

Eye Diagram/Jitter Analysis (software)

SDS6000Pro-EJ

I2S trigger & decode (software)

SDS6000Pro-I2S

MIL-STD-1553B trigger & decode (software)

SDS6000Pro-1553B

FlexRay trigger & decode (software)

SDS6000Pro-FlexRay

CAN FD trigger & decode (software)

SDS6000Pro-CANFDSENT trigger & decode (software)

SDS6000Pro-SENT

Manchester decode (software)

SDS6000Pro-Manch

500 MHz to 1 GHz bandwidth upgrade (software)

SDS6000-4BW10

1 GHz to 2 GHz bandwidth upgrade (software)

SDS6000-4BW20

STB3 demo signal source

STB3

High-speed active probe

SAP1000, SAP2500

High voltage probe

HPB4010High-speed differential probe

SAP2500D

High voltage differential probe

DPB1300/DPB4080/DPB5150/ DPB5150A/DPB5700/DPB5700A

Current probe

CPL5100/CP4020/CP4050/CP4070/CP4070A/CP5030/CP5030A/CP5150/CP5500

Rack Mount Kit

SDS6000-RMK

Bag

BAG-S2www.siglent.com 14

Digital Oscilloscope

SDS6000L
Low Profile Digital Storage Oscilloscope
Key Features• 8/4 analog channels + 1 external trigger. Designed for expansion. Combine multiple units for a high-speed acquisition system with up to
512 channels.
• Up to 2 GHz bandwidth with 5 GSa/s (10 GSa/s ESR) sample rate at each channel• Low background noise, supports 0.5 mV/div to 10 V/div vertical scales• SPO technology• Waveform capture rates up to 170,000 wfm/s (normal mode), and 750,000 wfm/s (sequence mode)• Supports 256-level intensity grading and color temperature display modes• 500 Mpts Record length in total for all 4 channels• Digital trigger system• Intelligent trigger: Edge, Slope, Pulse, Window, Runt, Interval, Dropout, Pattern, Qualified, Nth edge, Setup / hold, Delay and Video (HDTV
supported). Zone Trigger simplifies advanced triggering• Serial bus triggering and decoder, supports protocols I2C, SPI, UART, CAN, LIN, CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT and Manchester• Segmented acquisition (Sequence) mode, dividing the maximum record length into multiple segments (up to 80,000), according to trigger
conditions set by the user, with a very small dead time between segments to capture the qualifying event• History waveform record (History) function, the maximum recorded waveform length is 80,000 frames• Automatic measurements on 50+ parameters, supports statistics with histogram, track, trend, Gating measurement, and measurements on
Math, History, Memory and Ref• 4 Math traces (8 Mpts FFT, addition, subtraction, multiplication, division, integration, differential, square root, etc.), supports formula editor• Abundant data analysis functions such as Search, Navigate, Digital Voltmeter, Counter, Waveform Histogram, Power Analysis and Eye/Jitter
Analysis• 16 digital channels (optional)• 25 MHz function / arbitrary waveform generator, built-in multiple predefined waveforms• Interfaces include: 4x USB Hosts, USB Device (USBTMC), 1000M LAN (VXI-11/ Telnet/ Socket), micro SD card, Pass/Fail, Trigger Out, HDMI,
10MHz In, 10MHz Out• Built-in web server supports remote control over the LAN port using a web browser. Supports SCPI remote control commands. Supports
external mouse and keyboard
www.siglent.com 15

Digital Oscilloscope
Characteristics
Flexible Multi-channel High-speed Acquisition System• Standard sizes: 4 - channel models – 1U,8-channel models – 2U
• Multiple units are combined to create a high-speed acquisition system with up to 512 channels by being triggered with low-skew
synchronization signals from the 64-channel synchronization distributor SYN64• The host can access each unit over 1000M LAN. A complete SCPI command set as well as LabVIEW and IVI drivers are provided for easy dataacquisition. The LAN port is LXI compliant.
• Sample clocks are synchronized between all units in the test system by cascading the 10 MHz In and 10 MHz Out clocks in a daisy chain• High Waveform Update Rate

• Deep Record LengthWith a waveform update rate of up to 170,000 wfm/s, the oscilloscope

Using hardware-based Zoom technique and record length of up to 500

can easily capture unusual or low-probability events. In Sequence mode,Mpts, users can select a slower timebase without compromising thethe waveform capture rate can reach 750,000 wfm/s

sample rate, and then quickly zoom in to focus on the area of interest• Multiple Trigger Functions• Trigger Zone

Edge, Slope, Pulse, Video, Windows, Runt, Interval, Dropout, Pattern,

Trigger Zone is available for advanced triggering

Qualified, Nth edge, Setup/ hold, Delay, and serial trigger
www.siglent.com 16

Digital Oscilloscope• Search and Navigate• Serial Bus Decode

The oscilloscope can search events specified by the user in a frame.

Display the decoded characters through the events list. Bus protocol

Events flagged by the Search can be recalled automatically using

information can be quickly and intuitively displayed in tabular form. I2C,

Navigate. It can also navigate by time (delay position) and history frames

SPI, UART, CAN, LIN, CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT, andManchester are supported• Hardware-based High Speed Mask Test Functio

The oscilloscope utilizes a hardware-based Mask Test function,

Built-in Mask Editor application helps to create custom masks

performing up to 18,000 Pass / Fail decisions each second. It is easy togenerate user-defined test templates to provide trace mask comparisons,making it suitable for long-term signal monitoring or automatedproduction line testing• Eye / Jitter Analysis

Supports eye diagram and jitter analysis / measurement. It can automatically extract the embedded reference clock from serial data and create an eye diagram.
Measurement on multiple eye / jitter parameters is provided and mask testing of eye diagrams is supported

www.siglent.com 17

Digital Oscilloscope• Power Analysis (Optional)• Digital Channels / MSO (Optional)

The Power Analysis option provides a full suite of power measurements

Four analog channels plus 16 digital channels enable users to acquireand analysis, which greatly improves the measurement efficiency inand trigger the waveforms and then analyze the pattern, simultaneously

switching power supplies and power devices design

with one instrument

Specifications
Model

2 GHz

1 GHz

500 MHz8

SDS6208L

SDS6108L

SDS6058L

4

SDS6204L

SDS6104L

SDS6054L

Model

SDS6208L
SDS6204L

SDS6108L
SDS6104L

SDS6058L
SDS6054L

Channel

8/4 + EXT

Bandwidth

2 GHz

1 GHz

500 MHz

Sample rate (Max.)

5 GSa/s (10 GSa/s ESR) @ each channel

Channel

500 Mpts/ch (single-channel)
Memory depth (Max.)

250 Mpts/ch (dual-channel)
125 Mpts/ch (3 or 4 channels)

Waveform capture rate (Max.)
Vertical resolution
Trigger type
Serial trigger and decode
Measurement

Normal mode: 170,000 wfm/s;
Sequence mode: 750,000 wfm/s
8-bit
Up to 12-bit in ERES mode, equivalent to 16-bit Hi-Res mode
Edge, Slope, Pulse width, Window, Runt, Interval, Dropout, Pattern, Video, Qualified, Nth edge, Setup/hold, Delay, Serial
Standard: I2C, SPI, UART, CAN, LIN
Optional: CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT, Manchester (decode only)
50+ parameters, statistics, histogram, trend, and tracking supported
4 tracesMath

8 Mpts FFT, +, -, x, ÷, ∫dt, d/dt, √ , Identity, Negation, Absolute, Sign, ex, 10x, ln, lg, Interpolation, MaxHold, MinHold, ERES,
Average. Supports formula editor

Data analysis

Search, Navigate, History, Mask Test, Digital Voltmeter, Counter, Waveform Histogram, Power Analysis, Eye / Jitter Analysis

Digital channel (optional)

16-channel; maximum sample rate up to 1 GSa/s; record length up to 50 Mpts

Waveform generator (optional)

Single-channel external USB isolated waveform generator, frequency up to 25 MHz, 125 MSa/s sample rate, 16 kpts waveform
memoryHDMI (1280*800), USB 3.0 Host x2, USB 2.0 Host x2, USB 2.0 Device (USBTMC), 1000M LAN (SCPI over VXI-11, SCPI over

I/O

Telnet (port:5024), SCPI over Socket (port:5025), micro SD card, External trigger, Auxiliary output (TRIG OUT,PASS / FAIL), 10
MHz In, 10 MHz Out

Probe (Standard)

1x 500 MHz passive probe supplied for each channel

Multiple-channel Acquisition System
ChannelUp to 512

JitterWithin a unit: < 100 ps,rms

Skew
www.siglent.com 18

Between units:< 250 ps,rms

Without skew calibration, within a unit: < 100 ps
With skew calibration, within a unit: < 100 ps

Between units: < 500 ps
Between units: < 150 ps

Digital Oscilloscope
Ordering Information
Model

Description

SDS6208L

2 GHz bandwidth, 5 GSa/s sample rate, 8-bit, 500 Mpts memory depth, 8-channel

SDS6204L

2 GHz bandwidth, 5 GSa/s sample rate, 8-bit, 500 Mpts memory depth, 4-channel

SDS6108L

1 GHz bandwidth, 5 GSa/s sample rate, 8-bit, 500 Mpts memory depth, 8-channel

SDS6104L

1 GHz bandwidth, 5 GSa/s sample rate, 8-bit, 500 Mpts memory depth, 4-channel

SDS6058L

500 MHz bandwidth, 5 GSa/s sample rate, 8-bit, 500 Mpts memory depth, 8-channel

SDS6054L

500 MHz bandwidth, 5 GSa/s sample rate, 8-bit, 500 Mpts memory depth, 4-channel

Standard Accessories

Quantity

USB cable

1

Quick start

1

Passive probe

1/channel

Certificate of calibration

1

Wireless mouse

1

Power cord

1

Optional Accessories

Part No.

Waveform generator (software)

SDS6000L-FG

16 digital channels (software)

SDS6000L-16LA

16-channel logic probe

SPL2016

Power Analysis (software)

SDS6000L-PA

Power Analysis deskew fixture

DF2001A

Eye Diagram/Jitter Analysis (software)

SDS6000L-EJ

I2S trigger & decode (software)

SDS6000L-I2S

MIL-STD-1553B trigger & decode (software)

SDS6000L-1553B

FlexRay trigger & decode (software)

SDS6000L-FlexRay

CAN FD trigger & decode (software)

SDS6000L-CANFDSENT trigger & decode (software)

SDS6000L-SENT

Manchester decode (software)

SDS6000L-Manch

STB3 demo signal source

STB3

High-speed active probe

SAP1000, SAP2500

High voltage probe

HPB4010High-speed differential probe

SAP2500D

High voltage differential probe

DPB1300 / DPB4080 / DPB5150 / DPB5150A / DPB5700 / DPB5700A

Current probe
64-channel synchronization distributor

CPL5100 / CP4020 / CP4050 / CP4070 / CP4070A / CP6030 / CP6030A / CP6150/ CP6500
SYN64

www.siglent.com 19

Digital Oscilloscope

SDS5000X
Super Phosphor Oscilloscope
Key Features• 1 GHz, 500 MHz, 350 MHz models with real-time sampling rate up to 5 GSa/s• SPO technology• Waveform capture rate up to 110,000 wfm/s (normal mode), and 500,000 wfm/s (sequence mode)• Supports 256-level intensity grading and color temperature display modes• Record length up to 250 Mpts
• Digital trigger system• Intelligent trigger: Edge, Slope, Pulse Window, Runt, Interval, Dropout, Pattern, Qualified and Video (HDTV supported)2
2• Serial bus triggering and decoder, supports protocols I C, SPI, UART, CAN, LIN, CAN FD, FlexRay, I S and MIL-STD-1553B• Low background noise, supports 0.5 mV/div to 10 V/div voltage scales• Segmented acquisition (Sequence) mode, dividing the maximum record length into multiple segments (up to 100,000), according to trigger
conditions set by the user, with a very small dead time segments to capture the qualifying event• History waveform record (History) function, the maximum recorded waveform length is 100,000 frames• Automatic measurement function on more than 70 kinds of parameters, supports statistics, Gating measurement, Math measurement,
History measurement and Ref measurement• Math function (2 Mpts FFT, addition, subtraction, multiplication, division, integration, differential, square root)• Search and Navigate• Digital Voltmeter• High Speed hardware-based Average, ERES (Enhanced Resolution)• 16 digital channels (optional) with maximum waveform capture rate up to 1.25 GSa/s, record length up to 62.5 Mpts
• 25 MHz function / arbitrary waveform generator, built-in multiple predefined waveforms• Large 10.1” TFT-LCD display with 1024 * 600 resolution; Capacitive touch screen supports multi-touch gestures• Supports external mouse and keyboard• 10 types of one-button shortcuts• Multiple interfaces: USB Host, USB Device (USBTMC), LAN (VXI-11, telnet, socket, web), Pass / Fail, Trigger Out, 10 MHz In, 10 MHz Out,VGA output• Built-in web server supports remote control by the LAN port using a web browser• Supports SCPI remote control commands
www.siglent.com 20

Digital Oscilloscope
Characteristics• 10.1” TFT-LCD display with capacitive touch screen• Zone Trigger• 10.1” display with 1024*600 resolution• Capacitive touch screen, supporting multi-touch gestures,can move or scale the waveform traces quickly by finger-touch
movements, which greatly improves the operation efficiency.

Zone Trigger is available for advanced triggering• Record Length of up to 250 Mpts/ch• Multiple Trigger Functions

Using hardware-based Zoom technique and record length of up to 250

Edge, Slope, Pulse, Video, Windows, Runt, Interval, Dropout, Pattern,Mpts, users are able to select a slower timebase without compromising

Qualified, Nth edge, Setup/hold, Delay and serial trigger

the sampling rate, and then quickly zoom in to focus on the area ofinterest• Serial Bus Decode• Measurements of a Variety of Parameters

Display the decoded characters through the events list. Bus protocol
information can be quickly and intuitively displayed in tabular form. I2C,
SPI, UART, CAN, LIN, CAN FD, FlexRay and I2S and MIL-STD-1553B aresupported
Parameter measurements includes 3 categories: horizontal, verticaland CH delay providing more than 70 different types of measurements.
Measurements can be performed within a specified gate period.
Measurements on Math, Reference and History frames are supported
www.siglent.com 21

Digital Oscilloscope• Digital Voltmeter Function• Web control

4-digit voltmeter and 7-digit frequency counter. Any analog channel

With the new embedded web server, users can control the oscilloscope

can be selected as a source. Bar, Histogram and Trend diagrams arefrom a simple web page. This provides wonderful remote troubleshooting

supportedand monitoring capabilities.

• Bode Plot

• Power Analysis (Optional)

The SDS5000X can control the USB AWG module or a stand-alone SIGLENTThe Power Analysis option provides a full suite of power measurements

SDG generator, to scan the amplitude and phase frequency response of theand analysis, which greatly improve the measurement efficiency in

DUT, and display the data as a Bode Plot. This makes it possible to replaceswitching power supplies and power devices design.

expensive network analyzer in some applications.

• Digital Channels / MSO (Optional)• Built -in 25 MHz Function / Arbitrary Waveform
Generator (Optional)

Four analog channels plus 16 digital channels enable users to acquire

the SDS5000X can control the SAG1021I USB Function / Arbitrary

and trigger on the waveforms then analyze the pattern, simultaneously

waveform generator to output waveform with up to 25 MHz frequency

with one instrument

and ±3 V amplitude. Six basic waveforms plus multiple types of arbitrary
waveforms are built-in.

www.siglent.com 22

Digital Oscilloscope• Hardware-based Average and ERES Acquisition

NormalAverage• Complete Connectivity

ERES

Average and ERES (Enhanced Resolution) acquisition modes are

USB Host, USB Device (USBTMC), LAN (VXI-11, telnet, socket, web),

hardware-based, allowing the waveforms to be captured at a faster rate

Pass / Fail, Trigger Out, 10 MHz In / Out and VGA output

Specifications
Model

SDS5034X

SDS5054X

SDS5104X

Bandwidth

350 MHz

500 MHz

1 GHz

Sampling rate (Max.)

5 GSa/s (interleaving mode), 2.5 GSa/s (non-interleaving mode)

Analog channels

4 + EXT

Memory depth (Max.)

250 Mpts (interleaving mode), 125 Mpts (non-interleaving mode)

Waveform capture rate(Max.)

110,000 wfm/s (normal mode), 500,000 wfm/s (sequence mode)

Trigger type

Edge, Slope, Pulse width, Window, Runt, Interval, Dropout, Pattern, Video, Qualified

Serial trigger and decode

I C, SPI, UART, CAN, LIN, CAN FD, FlexRay, I S, MIL-STD-1553B

Digital channel (optional)

16-channel; maximum waveform capture rate up to 1.25 GSa/s; record length up to 62.5 Mpts

Waveform generator (optional)

Single channel, frequency up to 25 MHz, 125 MSa/s sample rate, 16 kpts waveform momory

I/O

USB Host, USB Device, LAN, Pass / Fail, Trigger Out, 10 MHz In, 10 MHz Out, VGA Output

Probe (standard)

1 probe supplied for each channel

Display

10.1” TFT-LCD with capacitive touch screen (1024*600)

22Ordering Information
Description

Model

1 GHz, 4 CH, 5 GSa/s (Max.)

SDS5104X

500 MHz, 4 CH, 5 GSa/s (Max.)

SDS5054X

350 MHz, 4 CH, 5 GSa/s (Max.)

SDS5034X

Standard Accessories
USB cable x1

Quick start x1

Certificate of calibration x1

Power cord x1

Passive probe x2 (2-ch model); x4 (4-ch model), SP2035A for 350 MHz models and SP3050A for 500 MHz / 1 GHz models
Optional Accessories
SDS-5000X-4BW05

350 MHz to 500 MHz bandwidth upgrade(4-ch model)

SDS-5000X-2BW05

350 MHz to 500 MHz bandwidth upgrade (2-ch model)

SDS-5000X-4BW10500 MHz to 1 GHz bandwidth upgrade (4-ch model)

SDS-5000X-2BW10500 MHz to 1 GHz bandwidth upgrade (2-ch model)

SDS-5000X-FG

Waveform generator software

SAG1021I

25 MHz USB function / arbitrary waveform generator

SDS-5000X-16LA

16 digital channels (software)

SPL2016

16-channel logic probe

SDS-5000X-I2S

I2S trigger & decode

SDS-5000X-CANFDCAN FD trigger & decode

SDS-5000X-FlexRay

FlexRay trigger & decode

SDS-5000X-1553B

MIL-STD-1553B trigger & decode

STB3

STB3 demo signal source

SAP1000

1 GHz active probe

HPB4010High voltage probe

CP4020 / CP4050 / CP4070 / CP4070A / CP5030 / CP5030A / CP5150 / CP5500

Current probe

DPB4080 / DPB5150 / DPB5150A / DPB5700 / DPB5700A

High voltage differential probe

www.siglent.com 23

Digital Oscilloscope

SDS3000X HD
Super Phosphor Oscilloscope
Key Features• 4 analog channels, up to 1 GHz bandwidth with up to 4GSa/s sample rate
• 12-bit ADC• Low background noise: 125 μVrms @ 1 GHz bandwidth• SPO technology• Waveform capture rates up to 200,000 wfm/s in Normal mode and 890,000 wfm/s in Sequence mode
• Supports 256-level intensity grading and color temperature display modes• Up to 400 Mpts/ch record length• Digital trigger system• Intelligent trigger: Edge, Slope, Pulse, Window, Runt, Interval, Dropout, Pattern, Qualified, Nth edge, Setup/hold, Delay and Video(HDTV supported). Zone Trigger simplifies advanced triggering• Serial bus triggering and decoder, supports protocols I2C, SPI, UART, CAN, LIN, CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT, Manchester and
ARINC429• Segmented acquisition (Sequence) mode, dividing the maximum record length into multiple segments (up to 80,000), according to trigger
conditions set by the user, with a very small dead time between segments to capture the qualifying event• History waveform record (History) function, the maximum recorded waveform length is 80,000 frames• Automatic measurements on 50+ parameters, supports statistics with histogram, track, trend, Gating measurement, and measurements on
Math, History and Memory traces• 4 Math traces (4 Mpts FFT, Filter, addition, subtraction, multiplication, division, integration, differential, square root, etc.), supports formula
editor• Abundant data analysis functions such as Search, Navigate, Digital Voltmeter, Counter, Waveform Histogram, Bode plot and Power Analysis• High Speed hardware-based Average, ERES; High Speed hardware-based Mask Test function, with Mask Editor tool for creating user-definedmasks• 16 digital channels (optional)• External 50 MHz waveform generator (optional)• Large 10.1” TFT-LCD display with 1024 * 600 resolution; Capacitive touch screen supports multi-touch gestures• Interfaces include: 2x USB 3.0 Host, 1x USB 2.0 Host , USB 3.0 Device (USBTMC), 1000M LAN ((VXI-11/Telnet/Socket/LXI)), External Triger
In, Aux Out (Pass/Fail, Trigger Out)• Built-in web server supports remote control over the LAN port using a web browser. Supports SCPI remote control commands. Supports
external mouse and keyboard
www.siglent.com 24

Digital Oscilloscope
Characteristics• 12-bit High Resolution

8-bit

12-bit12-bit resolution shows you more details and less noise on the waveform

Low noise floor: 125 µVrms at 1 GHz bandwidth

0.5% DC gain accuracy• Excellent User Interface and User Experience• 10.1” display with 1024x600 resolution• Capacitive touch screen, supporting multi-touch gestures,can move or scale the waveform traces quickly by finger-touch
movements, which greatly improves the operation efficiency• Built-in WebServer supports remote control on a web page
over LAN• Supports external mouse and keyboard• Multiple Trigger Functions• Zone Trigger

Edge, Slope, Pulse, Video, Windows, Runt, Interval, Dropout, Pattern,

Zone Trigger is available for advanced triggering. Combine spatial

Qualified, Nth edge, Setup/hold, Delay and serial trigger

triggering with common trigger modes to isolate signals of interest
www.siglent.com 25

Digital Oscilloscope• Deep Memory FFT• Serial Bus Decode

FFT supports up to 4 Mpts operation. This provides high-frequency

Display the decoded characters through the events list. Bus protocol

resolution with a fast refresh rate. The FFT function also supports a

information can be quickly and intuitively displayed in tabular form.

variety of window functions so that it can adapt to different spectrum

I2C, SPI, UART, CAN, LIN, CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT,

measurement needs. Three modes (Normal, Average, and Max hold) can

Manchester and ARINC429 are supported

satisfy different requirements for observing the power spectrum. Autopeak detection and markers are supported.

• Hardware-based High Speed Mask Test Function

The oscilloscope utilizes a hardware-based Mask Test function,

Built-in Mask Editor application helps to create custom masks

performing up to 28,000 Pass / Fail decisions each second. It is easy togenerate user-defined test templates to provide trace mask comparisons,making it suitable for long-term signal monitoring or automatedproduction line testing• Bode Plot

• Power Analysis (Optional)

The oscilloscope can control the SIGLENT isolated USB AWG module or a

The Power Analysis option provides a full suite of power measurements

stand-alone SIGLENT SDG generator, to scan the amplitude and phase-and analysis, which greatly improve the measurement efficiency infrequency response of the DUT, and display the data as a Bode Plot.

switching power supplies and power devices design

This makes it possible to replace expensive network analyzers in some
applications
www.siglent.com 26

Digital Oscilloscope• Digital Channels / MSO (Optional)• 50 MHz Function/Arbitrary Waveform Generator(Optional)

Four analog channels plus 16 digital channels enable users to acquire

The oscilloscope can control the SAG1021I waveform generator to output

and trigger the waveforms then analyze the pattern, simultaneously witha waveform with up to 50 MHz frequency and ±3 V amplitude. Six basic

one instrument

waveforms plus multiple types of arbitrary waveforms are built-in

Specifications
Model

SDS3104X HD

Analog channels

4 + EXT

Bandwidth
Sample rate (Max.)
Memory depth (Max.)
Waveform capture rate (Max.)

1 GHz(800 MHz in non-interleaving mode)

SDS3054X HD

SDS3034X HD

500 MHz

350 MHz

4 GSa/s (interleaving mode), 2 GSa/s (non-interleaving mode)
400 Mpts/ch (interleaving mode: single-channel), 200 Mpts/ch (interleaving mode: dual-channel),
100 Mpts/ch (non-interleaving mode)
Normal mode: 200,000 wfm/s;
Sequence mode: 890,000 wfm/s

Vertical resolution

12-bit. Up to 16-bit in ERES mode

Trigger type

Edge, Slope, Pulse width, Window, Runt, Interval, Dropout, Pattern, Video, Qualified, Nth edge, Setup/hold, Delay, Serial

Serial trigger and decode
Measurement

Standard: I2C, SPI, UART, CAN, LIN
Optional: CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT, Manchester (decode only), ARINC429
50+ parameters, statistics, histogram, trend, and track supported
4 traces

Math

4 Mpts FFT, +, -, x, ÷, ∫dt, d/dt, √, Identity, Negation, Absolute, Sign, ex, 10x, ln, lg, Interpolation, MaxHold, MinHold, ERES,
Average, Filter. Supports formula editor

Data analysis

Search, Navigate, History, Mask Test, Digital Voltmeter, Counter, Waveform Histogram, Bode plot, and Power Analysis

Digital channel (optional)

16-channel; maximum sample rate up to 1 GSa/s; record length up to 100 Mpts

Waveform generator (optional)

Single-channel SAG1021I, frequency up to 50 MHz, 125 MSa/s sample rate, 16 kpts waveform memory

I/O

USB 3.0 Host x2, USB 2.0 Host x1, USB 3.0 Device, 10M/100M/1000M LAN, External trigger, Auxiliary output (TRIG OUT,
PASS/FAIL)

Probe (Standard)

One 500 MHz passive probe supplied for each channel

Display

10.1 TFT-LCD with capacitive touch screen (1024*600)

www.siglent.com 27

Digital Oscilloscope
Ordering Information
Model

Description

SDS3104X HD

1 GHz, 4 GSa/s, 4-CH, 12-bit, 400 Mpts/ch memory depth, 10.1” capacitive touch screen

SDS3054X HD

500 MHz, 4 GSa/s, 4-CH, 12-bit, 400 Mpts/ch memory depth, 10.1” capacitive touch screen

SDS3034X HD

350 MHz, 4 GSa/s, 4-CH, 12-bit, 400 Mpts/ch memory depth, 10.1” capacitive touch screenStandard Accessories

Quantity

USB cable

1

Quick start

1

Passive probe (500 MHz)

1/channel

Certificate of calibration

1

Wireless mouse

1

Power cord

1

Optional Accessories

Part No.

USB isolated waveform generator

SAG1021I

16-channel logic probe

SPL2016

Power Analysis deskew fixture

DF2001A

STB3 demo signal source

STB3

USB-GPIB adapter

USB-GPIB

High-speed active probe

SAP1000, SAP2500

High voltage probe

HPB4010High-speed differential probe

SAP2500D

High voltage differential probe

DPB1300/DPB4080/DPB5150/ DPB5150A/DPB5700/DPB5700A

Current probe

CPL5100/CP4020/CP4050/CP4070/CP4070A CP6030/CP6030A/CP6150/
CP6500/SCP5030/SCP5030A/SCP5150/SCP5150A

Bag

BAG-S2

Options

Part No.

Waveform generator (software)

SDS3000HD-FG

16 digital channels (software)

SDS3000HD-16LA

Power Analysis (software)

SDS3000HD-PA

I2S trigger & decode (software)

SDS3000HD-I2S

MIL-STD-1553B trigger & decode (software)

SDS3000HD-1553B

FlexRay trigger & decode (software)

SDS3000HD-FlexRay

CAN FD trigger & decode (software)

SDS3000HD-CANFDSENT trigger & decode (software)

SDS3000HD-SENT

Manchester decode (software)

SDS3000HD-Manch

ARINC429 trigger & decode (software)

SDS3000HD-ARINC

350 MHz to 500 MHz bandwidth upgrade (software)

SDS3000HD-BW3T5

350 MHz to 1 GHz bandwidth upgrade (software)

SDS3000HD-BW3TA

500 MHz to 1 GHz bandwidth upgrade (software)

SDS3000HD-BW5TA

www.siglent.com 28

Digital Oscilloscope

SDS2000X HD
Super Phosphor Oscilloscope
Key Features• 12-bit High Resolution• 12-bit Analog-Digital Convertors with sample rate up to 2 GSa/s• Front ends with 70 μVrms noise floor @ 500 MHz bandwidth and 0.5% DC gain accuracy• 4 analog channels, up to 350 MHz bandwidth (upgradable to 500 MHz)• SPO technology• Waveform capture rate up to 100,000 wfm/s (normal mode), and 500,000 wfm/s (sequence mode)• Supports 256-level intensity grading and color temperature display modes• Up to 200 Mpts/ch record length• Digital trigger system• Intelligent trigger: Edge, Slope, Pulse, Window, Runt, Interval, Dropout, Pattern, Qualified, Nth edge, Setup/hold, Delay and Video(HDTV supported). Zone Trigger simplifies advanced triggering• Serial bus triggering and decoder, supports protocols I2C, SPI, UART, CAN, LIN, CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT and
Manchester• Segmented acquisition (Sequence) mode, dividing the maximum record length into multiple segments (up to 80,000), according to trigger
conditions set by the user, with a very small dead time between segments to capture the qualifying event• History waveform record (History) function, the maximum recorded waveform length is 80,000 frames• Automatic measurements on 50+ parameters, supports statistics with histogram, track, trend, Gating measurement, and measurements
on Math, History and Ref• 2 Math traces (2 Mpts FFT, addition, subtraction, multiplication, division, integration, differential, square root, etc.), supports formula editor• Abundant data analysis functions such as Search, Navigate, Digital Voltmeter, Counter, Waveform Histogram, Bode plot and Power Analysis• High Speed hardware-based Average, ERES; High Speed hardware-based Mask Test function, with Mask Editor tool for creatinguser-defined masks• 16 digital channels (optional)• Built-in 25 MHz waveform generator• Large 10.1” TFT-LCD display with 1024 * 600 resolution; Capacitive touch screen supports multi-touch gestures• Interfaces include: USB Hosts, USB Device (USBTMC), LAN (VXI-11/Telnet/Socket), Pass/Fail, Trigger Out• Built-in web server supports remote control over the LAN port using a web browser. Supports SCPI remote control commands. Supports
external mouse and keyboard
www.siglent.com 29

Digital Oscilloscope
Characteristics• 12-bit High Resolution

8-bit

12-bit12-bit resolution shows you more details and less noise on the waveform

Low noise floor: Only 70 µVrms at 500 MHz bandwidth

0.5% DC gain accuracy• Excellent User Interface and User Experience• 10.1” display with 1024x600 resolution• Capacitive touch screen, supporting multi-touch gestures,can move or scale the waveform traces quickly by finger-touch
movements, which greatly improves the operation efficiency• Built-in WebServer supports remote control on a web page
over LAN• Supports external mouse and keyboard• High Waveform Update Rate

• Deep Record LengthWith a waveform update rate of up to 100,000 wfm/s, the oscilloscope

Using hardware-based Zoom technique and record length of up to 200

can easily capture unusual or low-probability events. In Sequence mode,Mpts, users can select a slower timebase without compromising thethe waveform capture rate can reach 500,000 wfm/s

sample rate, and then quickly zoom in to focus on the area of interest

www.siglent.com 30

Digital Oscilloscope• Multiple Trigger Functions• Trigger Zone

Edge, Slope, Pulse, Video, Windows, Runt, Interval, Dropout, Pattern,

Trigger Zone is available for advanced triggering

Qualified, Nth edge, Setup/hold, Delay, and serial trigger• Advanced Math Function

In addition to the traditional (+, -, X, /) operations, FFT, integration,

Hardware-accelerated FFT supports up to 2 Mpts operation. This provides

differential, square root, and more are supported. Formula Editor ishigh-frequency resolution with a fast refresh rate. The FFT function alsoavailable for more complex operations. 2 math traces are available

supports a variety of window functions so that it can adapt to different
spectrum measurement needs. Three modes (Normal, Average, andMax hold) can satisfy different requirements for observing the power
spectrum. Auto peak detection and markers are supported• History Mode

• Sequence Mode

History function can record up to 80,000 frames of waveforms. The

Segmented memory collection will store the waveform into multiple

recording is executed automatically so that the customer can playback

memory segments (up to 80,000) and each segment will store athe history waveforms at any time to observe unusual events and quickly

triggered waveform as well the dead time information. The interval

locate the area of interest using the cursors or measurements. The failed

between segments can be as small as 2 μs. All of the segments can beframes of the Mask Test can be stored as history

played back using the History functionwww.siglent.com 31

Digital Oscilloscope• Search and Navigate• Serial Bus Decode

The oscilloscope can search events specified by the user in a frame.

Display the decoded characters through the events list. Bus protocol

Events flagged by the Search can be recalled automatically using

information can be quickly and intuitively displayed in tabular form. I2C,

Navigate. It can also navigate by time (delay position) and history frames

SPI, UART, CAN, LIN, CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT, andManchester are supported• Hardware-based High-Speed Mask Test Function

The oscilloscope utilizes a hardware-based Mask Test function,

Built-in Mask Editor application helps to create custom masks

performing up to 14,000 Pass / Fail decisions each second. It is easy togenerate user-defined test templates to provide trace mask comparisons,making it suitable for long-term signal monitoring or automatedproduction line testing• Bode Plot

• Power Analysis (Optional)

The oscilloscope can control the built-in waveform generator or a stand-

The Power Analysis option provides a full suite of power measurements

alone SIGLENT generator, to scan the amplitude and phase-frequency

and analysis, which greatly improve the measurement efficiency inresponse of the DUT, and display the data as a Bode Plot. This makes it

switching power supplies and power devices design

possible to replace expensive network analyzers in some applications
www.siglent.com 32

Digital Oscilloscope• Digital Channels / MSO (Optional)• Built-in 25 MHz Waveform Generator (Optional)

Four analog channels plus 16 digital channels enable users to acquire

The built-in waveform generator can output waveforms with up to 25

and trigger the waveforms then analyze the pattern, simultaneously with

MHz frequency and ±3 V amplitude. Six basic waveforms together with

one instrument

multiple types of predefined waveforms and as user-defined arbitrary
waveforms are supported• Complete Connectivity

USB Host 2.0 x3, USB Device 2.0 (USBTMC) , LAN (VXI-11/Telnet/Socket), Auxiliary output (Pass/Fail, Trigger Out), etc

Specifications
Model

SDS2354X HD

Analog channels

4 + EXT

Bandwidth

350 MHz,(upgradable to 500 MHz)SDS2204X HD

SDS2104X HD

200 MHz

100 MHz

Vertical resolution

12-bitSample rate (Max.)

2 GSa/s (interleaving mode), 1 GSa/s (non-interleaving mode)

Memory depth (Max.)

200 Mpts/ch (interleaving mode), 100 Mpts/ch (non-interleaving mode)

Waveform capture rate (Max.)
Trigger type
Serial trigger and decode
Measurement

Normal mode: 100,000 wfm/s;
Sequence mode: 500,000 wfm/s
Edge, Slope, Pulse width, Window, Runt, Interval, Dropout, Pattern, Video, Qualified, Nth edge, Setup/hold, Delay, Serial
Standard: I2C, SPI, UART, CAN, LIN
Optional: CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT, Manchester (decode only)
50+ parameters, statistics, histogram, trend, and track supported
2 tracesMath

2 Mpts FFT, +, -, x, ÷, ∫dt, d/dt, √, Identity, Negation, Absolute, Sign, ex, 10x, ln, lg, Interpolation, MaxHold, MinHold, ERES,
Average. Supports formula editor

Data analysis

Search, Navigate, History, Mask Test, Digital Voltmeter, Counter, Waveform Histogram, Bode plot, and Power Analysis

Digital channel (optional)

16-channel; maximum sample rate up to 500 MSa/s; record length up to 50 Mpts

Waveform generator (optional)

Single-channel built-in waveform generator, frequency up to 25 MHz, 125 MSa/s sample rate, 16 kpts waveform memory

I/O

USB 2.0 Host x3, USB 2.0 Device, 10 M / 100 M LAN, External trigger, Auxiliary output (TRIG OUT, PASS/FAIL)

Probe (Standard)

One 500 MHz passive probe supplied for each channel

Display

10.1 TFT-LCD with capacitive touch screen (1024*600)
www.siglent.com 33

Digital Oscilloscope
Ordering Information
Model

Description

SDS2354X HD

12-bit, 350 MHz, 2 GSa/s, 4-CH, 200 Mpts/ch memory depth, 10.1” capacitive touch screen

SDS2204X HD

12-bit, 200 MHz, 2 GSa/s, 4-CH, 200 Mpts/ch memory depth, 10.1” capacitive touch screen

SDS2104X HD

12-bit, 100 MHz, 2 GSa/s, 4-CH, 200 Mpts/ch memory depth, 10.1” capacitive touch screenStandard Accessories

Quantity

USB cable

1

Quick start

1

Passive probe ( 500 MHz)

1/channel

Certificate of calibration

1

Wireless mouse

1

Power cord

1

Optional Accessories

Part No.

Waveform generator (software)

SDS2000HD-FG

16 digital channels (software)

SDS2000HD-16LA

16-channel logic probe

SPL2016

Power Analysis (software)

SDS2000HD-PA

Power Analysis deskew fixture

DF2001AI2S trigger & decode (software)

SDS2000HD-I2S

MIL-STD-1553B trigger & decode (software)

SDS2000HD-1553B

FlexRay trigger & decode (software)

SDS2000HD-FlexRay

CAN FD trigger & decode (software)

SDS2000HD-CANFDSENT trigger & decode (software)

SDS2000HD-SENT

Manchester decode (software)

SDS2000HD-Manch

100 MHz to 200 MHz bandwidth upgrade (software)

SDS2000HD-BW1T2

100 MHz to 350 MHz bandwidth upgrade (software)

SDS2000HD-BW1T3

100 MHz to 500 MHz bandwidth upgrade (software)

SDS2000HD-BW1T5

200 MHz to 350 MHz bandwidth upgrade (software)

SDS2000HD-BW2T3

200 MHz to 500 MHz bandwidth upgrade (software)

SDS2000HD-BW2T5

350 MHz to 500 MHz bandwidth upgrade (software)

SDS2000HD-BW3T5

STB3 demo signal source

STB3

High voltage probe

HPB4010High voltage differential probe

DPB1300/DPB4080/DPB5150/DPB5150A/DPB5700/DPB5700A

Current probeBag

www.siglent.com 34

CPL5100/CP4020/CP4050/CP4070/CP4070A/CP6030/CP6030A/CP6150/
CP6500BAG-S2

Digital Oscilloscope

SDS2000X Plus
Super Phosphor Oscilloscope
Key Features• 350 MHz, 200 MHz, 100 MHz models with real-time sample rate up to 2 GSa/s. A 500 MHz bandwidth upgrade option is available for 350
MHz models.
• SPO technology
Waveform capture rates up to 120,000 wfm/s (normal mode) and 500,000 wfm/s (sequence mode)
Supports 256-level intensity grading and color temperature display modes
Record length up to 200 Mpts/ch, 400 Mpts in total for all 4 channels
Digital trigger system• 10-bit mode provides higher resolution and lower noise• Segmented acquisition (Sequence) mode, dividing the maximum record length into multiple segments (up to 90,000), according to trigger
conditions set by the user, with a very small dead time between segments to capture the qualifying event• History waveform record (History) function for up to 90,000 triggered waveforms (frames)• Automatic measurement function on 50+ parameters, supports statistics with histogram and trend• Two Math traces, support 2 Mpts FFT, +, -, x, ÷, d/dt, ∫dt, √, average, ERES, and formula editor• Abundant data processing and analysis functions such as Search, Navigate, Mask Test, Bode plot, Power Analysis (optional) and Counter• 16 digital channels (optional)• Built-in 50 MHz waveform generator (optional)• Large 10.1” TFT-LCD display with 1024x600 resolution; Capacitive touch screen supports multi-touch gestures• Multiple interfaces: USB Host, USB Device (USBTMC), LAN (VXI-11/Telnet/Socket), Pass/Fail, Trigger Out• Built-in web server supports remote control by the LAN port using a web browser; Supports SCPI remote control commands

www.siglent.com 35

Digital Oscilloscope
Characteristics• Competitive Front End Performance

500 MHz bandwidth (at 2 GSa/s sample rate with 500 MHz bandwidth

Low noise floor: Only 80 µV rms at 500 MHz bandwidth.

option).

• 10-bit Mode

8-bit

10-bit10-bit mode combined with Zoom shows you more details and less noise on the waveform.

• Trigger Zone

• Serial Bus Decode

Trigger Zone is available for advanced triggering.

In addition to the decoder lanes correlated to the waveform, bus protocol
information can be displayed in tabular form. I2C, SPI, UART, CAN, LIN,
CAN FD, FlexRay, I2S and MIL-STD-1553B are supported.

www.siglent.com 36

Digital Oscilloscope• Bode Plot

• Power Analysis (Optional)

The SDS2000X Plus can control the built-in waveform generator or

The Power Analysis option provides a full suite of power measurements

any stand-alone SIGLENT SDG device to scan the amplitude andand analysis, thus improving the efficiency of measurement in switchingphase response over frequency of passive or active circuits. The data

power supplies and power device designs.

is presented as Bode Plot. This makes it possible to replace expensive
network analyzers in less demanding applications.

• Digital Channels / MSO (Optional)• 50 MHz Built-in Waveform Generator (Optional)

Four analog channels plus 16 digital channels allow the acquisition and

The built-in waveform generator can output waveforms with up to 50

triggering of mixed waveforms with one instrument.

MHz frequency and ±3 V amplitude. Six basic waveforms together withmultiple types of predefined waveforms and as user defined arbitrary
waveforms are supported.

Specifications
Model

SDS2354X Plus

Analog channels

4 + EXT

Bandwidth

350 MHz,(upgradable to 500 MHz)SDS2204X Plus

2/4 + EXT
200 MHz

Sample rate (Max.)

2 GSa/s (interleaving mode), 1 GSa/s (non-interleaving mode)

Memory depth (Max.)

200 Mpts/ch (interleaving mode), 100 Mpts/ch (non-interleaving mode)

Waveform capture rate (Max.)

Sequence mode: 500,000 wfm/s
8-bit. 10-bit mode (with typical 100 MHz bandwidth)

Trigger type

Edge, Slope, Pulse, Window, Runt, Interval, Dropout, Pattern, Video and Serial

Measurement
MathData processing and analysis tools

100 MHz

Normal mode: 120,000 wfm/s;

Vertical resolution

Serial trigger and decode

SDS2104X Plus
SDS2102X Plus

Standard: I2C, SPI, UART, CAN, LIN
Optional: CAN FD, FlexRay, I2S, MIL-STD-1553B
More than 50 parameters, supports statistics with histogram and trend
2 traces
2 Mpts FFT, +, -, x, ÷, d/dt, ∫dt, √, average, ERES, and formula editor
Search, Navigate, History, Mask test, Bode plot, Power Analysis (optional) and Counter
www.siglent.com 37

Digital Oscilloscope
Digital channel (optional)

16-channel; maximum sample rate up to 500 MSa/s; record length up to 50 Mpts/ch

Waveform generator (optional)

Single channel, frequency up to 50 MHz, 125 MSa/s sample rate, 16 kpts waveform memory

InterfaceUSB 2.0 Host x2, USB 2.0 Device, LAN, External trigger, Auxiliary output (TRIG OUT, PASS/FAIL)

Probe (standard)
Display

SP2035A, 350 MHz, 1 probe supplied foreach channel

PP215, 200 MHz, 1 probe supplied for each channel

10.1” TFT-LCD with capacitive touch screen (1024x600)

Ordering Information
Model

Description

SDS2354X Plus

350 MHz, 4-ch, 2 GSa/s (Max.), 200 Mpts, 10.1”touch screen

SDS2204X Plus

200 MHz, 4-ch, 2 GSa/s (Max.), 200 Mpts, 10.1”touch screen

SDS2104X Plus

100 MHz, 4-ch, 2 GSa/s (Max.), 200 Mpts, 10.1”touch screen

SDS2102X Plus

100 MHz, 2-ch, 2 GSa/s (Max.), 200 Mpts, 10.1”touch screenStandard Accessories

Quantity

USB cable

1

Quick start

1

Passive probe

x2 (2-ch model); x4 (4-ch model)

Certificate of calibration

1

Power cord

1

Optional Accessories

Part Number

Waveform generator option (software)

SDS2000XP-FG

16 digital channels (software)

SDS2000XP-16LA

16-channel logic probe

SPL2016

Power Analysis (software)

SDS2000XP-PA

Power Analysis deskew fixture

DF2001AI2S trigger & decode (software)

SDS2000XP-I2S

MIL-STD-1553B trigger & decode (software)

SDS2000XP-1553B

FlexRay trigger & decode (software)

SDS2000XP-FlexRay

CAN FD trigger & decode (software)

SDS2000XP-CANFD

100 MHz to 200 MHz bandwidth upgrade (4-ch model) (software)

SDS2000XP-4BW02

200 MHz to 350 MHz bandwidth upgrade (4-ch model) (software)

SDS2000XP-4BW03

350 MHz to 500 MHz bandwidth upgrade (4-ch model) (software)

SDS2000XP-4BW05

100 MHz to 350 MHz bandwidth upgrade (2-ch model) (software)

SDS2000XP-2BW03

STB3 demo signal source

STB3

High voltage probe

HPB4010High voltage differential probe

DPB1300/DPB4080/DPB5150/DPB5150A/DPB5700/DPB5700A

Current probeBag

www.siglent.com 38

CPL5100/CP4020/CP4050/CP4070/CP4070A/CP5030/CP5030A/CP5150/
CP5500BAG-S2

Digital Oscilloscope

SDS2000X-E
Super Phosphor Oscilloscope
Key Features• 200 MHz, 350 MHz bandwidth models• Real-time sampling rate up to 2 GSa/s (1 GSa/s per channel, if both channels active)• Record length up to 28 Mpts
• Intelligent triggers: Edge, Slope, Pulse Width, Window, Runt, Interval, Time out (Dropout), Pattern
2• Serial bus triggering and decoding (standard), supports protocols I C, SPI, UART, CAN, LIN• Low background noise with voltage scales from 500 μV/div to 10 V/div• 10 types of one-button shortcuts, supports Auto Setup, Default, Cursors, Measure, Roll, History, Display/Persist, Clear Sweep, Zoom and Print• History waveform record (history) function (maximum recorded waveform length is 80,000 frames)• 1 Mpts FFT• Math and measurement functions use all sampled data points in memory (up to 28 Mpts)• Math functions (FFT, addition, subtraction, multiplication, division, integration, differential, square root)• Large 7 inch TFT -LCD display with 800 * 480 resolution• Supports Multi-language display and embedded online help

www.siglent.com 39

Digital Oscilloscope
Characteristics• Maximum sample rate of 2 GSa/s, record Lengthof up to 28 Mpts

• Waveform Capture Rate up to 400,000 wfm/s

Using hardware-based Zoom technologies and max record length of up toWith a waveform capture rate of up to 400,000 wfm/s (sequence mode),

28 Mpts, users are able to oversample to capture for longer time periods

the oscilloscope can easily capture the unusual or low-probability events.

at higher resolution and use the zoom feature to see more details withineach signal.

• Serial Bus Decoding Function (Standard)• History Waveforms (History) Mode andSegmented Acquisition (Sequence)

SDS2000X-E displays the decoding through the events list. Bus protocol

Playback the latest triggered events using the history function.

information can be quickly and intuitively displayed in a tabular format.

Segmented memory collection will store trigger events into multiple(Up to 80,000) memory segments, each segment will store triggered
waveforms and timestamp of each frame.

• True measurement to 28 Mpts

• 1 Mpts FFTSDS2000X-E can apply automatic measurements on all sampled data

The new math co-processor enables FFT analysis of incoming signals

points up to 28 Mpts. This ensures the accuracy of measurements while

using up to 1 million samples per waveform. This provides high frequency

the math co-processor decreases measurement time and increases ease-resolution with a fast refresh rate. The FFT function also supports aof-use.

variety of window functions so that it can adapt to different spectrum
measurement needs.

www.siglent.com 40

Digital Oscilloscope• Hardware-Based High Speed Pass/Fail function• 16 Digital Channels/MSO (option)

The SDS2000X-E utilizes a hardware-based Pass/Fail function, performing

16 digital channels enables users to acquire and trigger on digital input

up to 40,000 Pass / Fail decisions each second. Easily generate user

channels and view both digital and analog waveforms simultaneously

defined test templates provide trace mask comparison making it suitablewith one instrument.

for long-term signal monitoring or automated production line testing.

• USB 25 MHz AWG Module (option)• Search and Navigate

The optional 25 MHz function/arbitrary waveform generator is operated

The SDS2000X-E can search events specified by the user in a frame. It

from the USB host connection. Functions include Sine, Square, Ramp,

can also navigate by time (delay position) and historical frames.

Pulse, Noise, DC and 45 additional built-in waveforms. The arbitrary
waveforms can be accessed and edited by the SIGLENT EasyWave PC
software.

• Bode Plot

• Web control

SDS2000X-E can control the USB AWG module or an independent

With the new embedded web server, users can control the SDS2000X-E fromSIGLENT SDG instrument, scan a circuits amplitude and phase frequency

a simple web page. This provides wonderful remote troubleshooting andresponse, and display the data as a Bode Plot. It can also show the resultmonitoring capabilities. The web page has PC and mobile styles that include

lists, and export the data to a USB disk.

an embedded virtual control panel.

www.siglent.com 41

Digital Oscilloscope
Models and key Specification
Model

SDS2202X-E

SDS2352X-E

Bandwidth

200 MHz

350 MHz

Sample Rate (Max.)

2 GSa/s

Channels

2+EXT

Memory Depth (Max.)

14 Mpts/CH (not interleave mode)
28 Mpts/CH (interleave mode)

Waveform Capture Rate (Max.)

110,000 wfm/s (normal mode), 400,000 wfm/s (sequence mode)

Trigger Type

Edge, Slope, Pulse Width, Window, Runt, Interval, Dropout, Pattern, Video

Serial Trigger and decoder (Standard)

I C, SPI, UART, CAN, LIN

16 Digital Channels (option)

Maximum waveform capture rate up to 1GSa/s, Record length up to 14 Mpts/CH

USB AWG module (option)

One channel, 25 MHz, sample rate of 125 MHz, 16 kpts waveform memory sample size

Bode plot

2

Minimum start frequency of 10 Hz, minimum scan bandwidth of 500 Hz, maximum scan bandwidth of 120 MHz(dependent on Oscilloscope and AWG bandwidth), 500 maximum scan frequency points

I/O

USB Host, USB Device, LAN, Pass/Fail, Trigger Out, Sbus (Siglent MSO)

Probe (Std)2 pcs passive probe PP215

Display

7 inch TFT–LCD (800 x 480 pixels)

WeightWithout package 2.6 Kg; With package 3.8 Kg

2 pcs passive probe SP2035

Ordering Information
SDS2000X-E Series Digital Oscilloscope
Product NameSDS2202X-E 200 MHz
SDS2352X-E 350 MHz
USB Cable -1
Quick Start -1

Standard Accessories

Passive Probe -2
Certification of Calibration -1
Power Cord -1

Optional Accessories

16 Channels MSO Software

SDS2000X-E-16LA

16 Channels Logic Analyzer

SLA1016AWG Software

SDS2000X-E-FG

USB AWG Module Hardware

SAG1021I

STB Demo Source

STB-3

High Voltage Probe

HPB4010

Current Probeswww.siglent.com 42

CP4020/CP4050/CP4070/CP4070A/CP5030/CP5030A/
CP5150/CP5500

Differential Probes

DPB4080/DPB5150/DPB5150A/DPB5700/DPB5700A

Rack Mount

SDS1X-E-RMKDigital Oscilloscope

SDS1000X HD
Super Phosphor Oscilloscope
Key Features• 12-bit High Resolution• 12-bit Analog-Digital Convertors with sample rate up to 2 GSa/s• Front ends with 70 μVrms noise floor @ 200 MHz bandwidth• 2/4 analog channels, up to 200 MHz bandwidth• SPO technology• Waveform capture rate up to 120,000 wfm/s (normal mode), and 500,000 wfm/s (sequence mode)• Supports 256-level intensity grading and color temperature display modes• Up to 100 Mpts record length• Digital trigger system• Intelligent trigger: Edge, Slope, Pulse width, Window, Runt, Interval, Dropout, Pattern, Video (HDTV supported) , Qualified, Nth edge, Delay,
Setup/Hold time.
• Serial bus triggering and decoder, supports protocols I2C, SPI, UART, CAN, LIN, CAN FD(decode only), FlexRay(decode only)• Segmented acquisition (Sequence) mode, dividing the maximum record length into multiple segments (up to 80,000), according to trigger
conditions set by the user, with a very small dead time between segments to capture the qualifying event• History waveform record (History) function, the maximum recorded waveform length is 80,000 frames• Automatic measurements on 50+ parameters, supports statistics with histogram, track, trend, Gating measurement, and measurements on
Math, History and Ref• 4 Math traces (2 Mpts FFT, addition, subtraction, multiplication, division, integration, differential, square root, etc.), supports formula editor• Abundant data analysis functions such as Search, Navigate, Counter, Bode plot and Power Analysis• High Speed hardware-based Mask Test function, with Mask Editor tool for creating user-defined masks• 16 digital channels (optional)• 25 MHz waveform generator(optional)• Large 10.1” TFT-LCD display with 1024 * 600 resolution; Capacitive touch screen supports multi-touch gestures• Interfaces include: USB Hosts, USB Device (USBTMC), LAN (VXI-11/Telnet/Socket), Pass/Fail, Trigger Out• Built-in web server supports remote control over the LAN port using a web browser. Supports SCPI remote control commands. Supports
external mouse and keyboard

www.siglent.com 43

Digital Oscilloscope
Characteristics• 12-bit High Resolution

8-bit

12-bitVertical & Horizontal Zoom along with a large memory depth make the most out of 12-bit ADC resolution. Engineers can observe waveform overall and details
simultaneously.

Low noise floor: Only 70 µVrms at 200 MHz bandwidth

0.5% DC gain accuracy• Excellent User Interface and User Experience• 10.1” display with 1024x600 resolution• Capacitive touch screen, supporting multi-touch gestures,can moveor scale the waveform traces quickly by finger-touch movements,
which greatly improves the operation efficiency• Built-in WebServer supports remote control on a web page overLAN
• Supports external mouse and keyboard• Serial Bus Decode• Advanced Math Function

Display the decoded characters through the events list. Bus protocol

Hardware-accelerated FFT supports up to 2 Mpts operation. This provides
high-frequency resolution with a fast refresh rate. The FFT function also
supports a variety of window functions so that it can adapt to different
spectrum measurement needs. Three modes (Normal, Average, andMax hold) can satisfy different requirements for observing the power
spectrum. Auto peak detection and markers are supported.

information can be quickly and intuitively displayed in tabular form. I2C,
SPI, UART, CAN, LIN, CAN FD, FlexRay are supported

www.siglent.com 44

Digital Oscilloscope• Hardware-based High-Speed Mask Test Function

The oscilloscope utilizes a hardware-based Mask Test function,

Built-in Mask Editor application helps to create custom masks

performing up to 80,000 Pass / Fail decisions each second. It is easy togenerate user-defined test templates to provide trace mask comparisons,making it suitable for long-term signal monitoring or automatedproduction line testing• Digital Channels / MSO (Optional)• USB AWG module (Optional)

Four analog channels plus 16 digital channels enable users to acquire

The USB waveform generator can output waveforms with up to 25

and trigger the waveforms then analyze the pattern, simultaneously with

MHz frequency and ±3 V amplitude. Six basic waveforms together with

one instrument

multiple types of predefined waveforms and as user-defined arbitrary
waveforms are supported• Bode Plot

• Power Analysis (Optional)

SDS1000X HD can control the USB AWG module or control an

The Power Analysis option provides a full suite of power measurements

independent SIGLENT SDG instrument, scan a devices amplitude andand analysis, which greatly improve the measurement efficiency inphase frequency response, and display the data as a Bode Plot. Thereswitching power supplies and power devices designis also a Vari-level Mode for accurately measuring Power Supply Control
Loop Response (PSRR)
www.siglent.com 45

Digital Oscilloscope
Specifications
Model

SDS1104X HD
SDS1102X HD

SDS1204X HD
SDS1202X HD

Analog channels

4 + EXT (4CH Series: SDSxxx4X HD), 2 + EXT (2CH Series: SDSxxx2X HD)

Bandwidth

100 MHz

Vertical resolution

12-bitSample rate (Max.)One channel mode: 2 GSa/s, Two channel mode: 1 GSa/s, Four channel mode: 500 MSa/s

Memory depth (Max.)One channel mode: 100 Mpts/ch, Two channel mode: 50 Mpts/ch, Four channel mode: 25 Mpts/ch

Waveform capture rate (Max.)

Normal mode: 120,000 wfm/s; Sequence mode: 500,000 wfm/s

Trigger type

Edge, Slope, Pulse width, Window, Runt, Interval, Dropout, Pattern, Video, Qualified, Nth edge, Delay, Setup/Hold time, Serial

200 MHz

Serial trigger and decode(Standard) I2C, SPI, UART, CAN, LIN, CAN FD (Decode Only), FlexRay (Decode Only)
Measurement

50+ parameters, statistics, histogram, trend, and track supported
4 tracesMath

2 Mpts FFT, Filter, +, -, x, ÷, ∫dt, d/dt, √, Identity, Negation, Absolute, Sign, ex, 10x, ln, lg, Interpolation, MaxHold, MinHold,
ERES, Average. Supports formula editor

Data analysis

Search, Navigate, History, Mask Test, Counter, Bode plot, and Power Analysis

Digital channel (optional)

16-channel; maximum sample rate up to 1 GSa/s; record length up to 10 Mpts

USB AWG module (option)

One channel, 25 MHz, sample rate of 125 MHz, wave length of 16 kpts, isolated output

I/O

USB 2.0 Host x3, USB 2.0 Device, 10 M / 100 M LAN, External trigger, Auxiliary output (TRIG OUT, PASS/FAIL),
SBUS (Siglent MSO)

Probe (Standard)

Passive probe PP510 for each channel

Display

10.1 TFT-LCD with capacitive touch screen (1024*600)

Passive probe PP215 for each channel

Ordering Information
Model

Description

SDS1204X HD

200 MHz, 2 GSa/s, 4CH

SDS1104X HD

100 MHz, 2 GSa/s, 4CH

SDS1202X HD

200 MHz, 2 GSa/s, 2CH

SDS1102X HD

100 MHz, 2 GSa/s, 2CH

Standard Accessories

Quantity

USB cable

1

Quick start

1

Passive probe

1/channel

Certificate of calibration

1

Power cord

1

Optional Accessories

Part No.

AWG Software

SDS1000XHD-FG

USB Isolated AWG Module Hardware

SAG1021I

16 Channels MSO Software

SDS1000XHD-16LA

16 Channels Logic Analyzer

SLA1016

Power Analysis Software

SDS1000XHD-PA

Power Analysis deskew fixture

DF2001A

www.siglent.com 46

Digital Oscilloscope

SDS800X HD
Super Phosphor Oscilloscope
Key Features• 12-bit High Resolution• 12-bit Analog-Digital Convertors with sample rate up to 2 GSa/s• Front ends with 70 μVrms noise floor @ 200 MHz bandwidth• 2/4 analog channels, up to 200 MHz bandwidth• SPO technology• Waveform capture rate up to 120,000 wfm/s (normal mode), and 500,000 wfm/s (sequence mode)• Supports 256-level intensity grading and color temperature display modes• Up to 100 Mpts record length• Digital trigger system• Intelligent trigger: Edge, Slope, Pulse width, Window, Runt, Interval, Dropout, Pattern, Video (HDTV supported) , Qualified, Nth edge, Delay,
Setup/Hold time.
• Serial bus triggering and decoder, supports protocols I2C, SPI, UART, CAN, LIN• Segmented acquisition (Sequence) mode, dividing the maximum record length into multiple segments (up to 80,000), according to trigger
conditions set by the user, with a very small dead time between segments to capture the qualifying event• History waveform record (History) function, the maximum recorded waveform length is 80,000 frames• Automatic measurements on 50+ parameters, supports statistics with histogram, track, trend, Gating measurement, and measurements on
Math, History and Ref• 4 Math traces (2 Mpts FFT, addition, subtraction, multiplication, division, integration, differential, square root, etc.), supports formula editor• Abundant data analysis functions such as Search, Navigate, Counter, Bode plot and Power Analysis• High Speed hardware-based Mask Test function, with Mask Editor tool for creating user-defined masks• 16 digital channels (optional)• 25 MHz waveform generator(optional)• 7” TFT-LCD display with 1024 * 600 resolution; Capacitive touch screen supports multi-touch gestures• Interfaces include: USB Hosts, USB Device (USBTMC), LAN (VXI-11/Telnet/Socket), Pass/Fail, Trigger Out• Built-in web server supports remote control over the LAN port using a web browser. Supports SCPI remote control commands. Supports
external mouse and keyboard. Supports NTP

www.siglent.com 47

Digital Oscilloscope
Characteristics• 12-bit High Resolution

8-bit

12-bitVertical & Horizontal Zoom along with a large memory depth make the most out of 12-bit ADC resolution. Engineers can observe waveform overall and details
simultaneously.

Low noise floor: Only 70 µVrms at 200 MHz bandwidth

0.5% DC gain accuracy• Excellent User Interface and User Experience• 7” display with 1024x600 resolution• Capacitive touch screen, supporting multi-touch gestures,can moveor scale the waveform traces quickly by finger-touch movements,
which greatly improves the operation efficiency• Built-in WebServer supports remote control on a web page overLAN
• Supports external mouse and keyboard• History Mode

• Advanced Math Function

History function can record up to 80,000 frames of waveforms. The
recording is executed automatically so that the customer can playback
the history waveforms at any time to observe unusual events and quickly
locate the area of interest using the cursors or measurements

Hardware-accelerated FFT supports up to 2 Mpts operation. This provides
high-frequency resolution with a fast refresh rate. The FFT function also
supports a variety of window functions so that it can adapt to different
spectrum measurement needs. Three modes (Normal, Average, andMax hold) can satisfy different requirements for observing the power
spectrum. Auto peak detection and markers are supported

www.siglent.com 48

Digital Oscilloscope• Sequence Mode

• Serial Bus Decode

Segmented memory collection will store the waveform into multiple

Display the decoded characters through the events list. Bus protocol

memory segments (up to 80,000) and each segment will store ainformation can be quickly and intuitively displayed in tabular form. I2C,

triggered waveform as well the dead time information. The interval

SPI, UART, CAN, LIN are supported

between segments can be as small as 2 μs. All of the segments can beplayed back using the History function• Hardware-based High-Speed Mask Test Function

The oscilloscope utilizes a hardware-based Mask Test function,

Built-in Mask Editor application helps to create custom masks

performing up to 80,000 Pass / Fail decisions each second. It is easy togenerate user-defined test templates to provide trace mask comparisons,making it suitable for long-term signal monitoring or automatedproduction line testing• Bode Plot

• Power Analysis (Optional)

SDS800X HD can control the USB AWG module or control an independent

The Power Analysis option provides a full suite of power measurements

SIGLENT SDG instrument, scan a devices amplitude and phase frequency

and analysis, which greatly improve the measurement efficiency inresponse, and display the data as a Bode Plot. There is also a Vari-levelswitching power supplies and power devices design

Mode for accurately measuring Power Supply Control Loop Response(PSRR)
www.siglent.com 49

Digital Oscilloscope• Digital Channels / MSO (Optional)• USB AWG module (Optional)

Four analog channels plus 16 digital channels enable users to acquire

The USB waveform generator can output waveforms with up to 25

and trigger the waveforms then analyze the pattern, simultaneously with

MHz frequency and ±3 V amplitude. Six basic waveforms together with

one instrument

multiple types of predefined waveforms and as user-defined arbitrary
waveforms are supported

Specifications
Model

SDS804X HD
SDS802X HD

Analog channels

4 (4CH Series: SDS804X HD, SDS814X HD, SDS824X HD), 2 (2CH Series: SDS802X HD, SDS812X HD, SDS822X HD)

Bandwidth

70 MHzVertical resolution

12-bitSample rate (Max.)One channel mode: 2 GSa/s, Two channel mode: 1 GSa/s, Four channel mode: 500 MSa/s

Memory depth (Max.)

Waveform capture rate (Max.)
Trigger type

SDS814X HD
SDS812X HD
100 MHz

SDS824X HD
SDS822X HD
200 MHzOne channel mode: 50 Mpts/ch,

One channel mode: 100 Mpts/ch,

Two channel mode: 25 Mpts/ch,

Two channel mode: 50 Mpts/ch,

Four channel mode: 10 Mpts/ch

Four channel mode: 25 Mpts/ch

Normal mode: 80,000 wfm/s;

Normal mode: 120,000 wfm/s;

Sequence mode: 500,000 wfm/s

Sequence mode: 500,000 wfm/s

Edge, Slope, Pulse width, Window, Runt, Interval, Dropout, Pattern, Video, Qualified, Nth edge, Delay, Setup/Hold time, Serial

Serial trigger and decode(Standard) I2C, SPI, UART, CAN, LIN
Measurement

50+ parameters, statistics, histogram, trend, and track supported
4 tracesMath

2 Mpts FFT, Filter, +, -, x, ÷, ∫dt, d/dt, √, Identity, Negation, Absolute, Sign, ex, 10x, ln, lg, Interpolation, MaxHold, MinHold,
ERES, Average. Supports formula editor

Data analysis

Search, Navigate, History, Mask Test, Counter, Bode plot, and Power Analysis

Digital channel (optional)

16-channel; maximum sample rate up to 1 GSa/s; record length up to 10 Mpts

USB AWG module (option)

One channel, 25 MHz, sample rate of 125 MHz, wave length of 16 kpts, isolated output

I/O

USB 2.0 Host x2, USB 2.0 Device, 10 M / 100 M LAN, Auxiliary output (TRIG OUT, PASS/FAIL), SBUS (Siglent MSO)

Probe (Standard)

Passive probe PB470 for each channel

Display

7 TFT-LCD with capacitive touch screen (1024*600)

www.siglent.com 50

Passive probe PP510 for each channel

Passive probe PP215 for each channel

Digital Oscilloscope
Ordering Information
Model

Description

SDS824X HD

200 MHz, 2 GSa/s, 4CH

SDS814X HD

100 MHz, 2 GSa/s, 4CH

SDS804X HD

70 MHz, 2 GSa/s, 4CH

SDS822X HD

200 MHz, 2 GSa/s, 2CH

SDS812X HD

100 MHz, 2 GSa/s, 2CH

SDS802X HD

70 MHz, 2 GSa/s, 2CH

Standard Accessories

Quantity

USB cable

1

Quick start

1

Passive probe

1/channel

Certificate of calibration

1

Power cord

1

Optional Accessories

Part No.

AWG Software

SDS800XHD-FG

USB Isolated AWG Module Hardware

SAG1021I

16 Channels MSO Software

SDS800XHD-16LA

16 Channels Logic Analyzer

SLA1016

Power Analysis Software

SDS800XHD-PA

Power Analysis deskew fixture

DF2001A

www.siglent.com 51

Digital Oscilloscope

SDS1000X-E Super Phosphor Oscilloscope
Key Features• Two channel series have one 1 GSa/s ADC, four channel series have two 1 GSa/s ADCs. When all channels are enabled, each channel has amaximum sample rate of 500 MSa/s. When a single channel per ADC is active, it has sample rate of 1 GSa/s• The newest generation of SPO technology• Waveform capture rate up to 100,000 wfm/s (normal mode), and 400,000 wfm/s (sequence mode)• Supports 256-level intensity grading and color display modes• Record length up to 14 Mpts
• Digital trigger system• Intelligent trigger: Edge, Slope, Pulse Width, Window, Runt, Interval, Time out (Dropout), Pattern• Serial bus triggering and decoding (Standard), supports protocols I2C, SPI, UART, RS232, CAN, LIN• Segmented acquisition (Sequence) mode, divides the maximum record length into multiple segments (up to 80,000), according to trigger
conditions set by the user, with a very small dead time segment to capture the qualifying event• 1 Mpts FFT• Math and measurement functions use all sampled data points (up to 14 Mpts)• MSO, 16 digital channels (four channel series only, optional)• Search and navigate (four channel series only)• USB AWG module (four channel series only, optional)

www.siglent.com 52

Digital Oscilloscope
Function & Characteristics• When all channels are enabled, each channel has a maximum sample rate of 500 MSa/s. When a single
channel per pair is active, that channel has sample rate of 1 GSa/s

The four channel series has two 1 GSa/s ADC chips (channel 1 and 2 share one, channel 3 and 4 share another), so that each channel can achieve sample rates up
to 500 MSa/s and work on bandwidths of 200 MHz when all channels are enabled.

• Record Length of Up to 14 Mpts (single channel/• Waveform Capture Rate Up to 400,000 wfm/s

pair mode), 7 Mpts/CH (two channels/pair mode)

Using hardware-based Zoom technologies and max record length of up toWith a waveform capture rate of up to 400,000 wfm/s (sequence mode),

14 Mpts, users are able to oversample to capture for longer time periods

the oscilloscope can easily capture the unusual or low-probability events.

at higher resolution and use the zoom feature to see more details withineach signal.

• 256 -Level Intensity Grading and Color Temperature Display

SPO display technology provides for fast refresh rates. The resulting

The color temperature display is similar to the intensity-graded trace

intensity-graded trace is brighter for events that occur with morefunction, but the trace occurrence is represented by different colors (color

frequency and dims when the events occur with less frequency.

“temperature”) as opposed to changes in the intensity of one color. Red
colors represents the more frequent events, while blue is used to markpoints that occur less frequently.
www.siglent.com 53

Digital Oscilloscope• Search and Navigate (four channel series only)

The SDS1000X-E can search events specified by the user in a frame. It can also navigate by time (delay position) and historical frames.

• 16 Digital Channels/MSO (four channel series
only, optional)• USB 25 MHz AWG Module (four channel series
only, optional)

16 digital channels enables users to acquire and trigger on the

The four channel series supports a USB 25 MHz function/arbitrary

waveforms then analyze the pattern, simultaneously with one instrument.

waveform generator that is operated from the USB host connection.
Functions include Sine, Square, Ramp, Pulse, Noise, DC and 45 built-in
waveforms. The arbitrary waveforms can be accessed and edited by the
SIGLENT EasyWave PC software.

• 7 inch TFT-LCD display and 10 one-button menus

Front panel of the four channel series• 7 -inch TFT -LCD display with 800 * 480 resolution

Front panel of the two channel series• Most commonly used functions are accessible using 10 different one-button operation keys: Auto Setup, Default, Cursor, Measure, Roll, History, Persist, Clear
Sweep, Zoom, Print

www.siglent.com 54

Digital Oscilloscope
Models and key Specification
Model

SDS1104X-E

SDS1204X -E
SDS1202X-E

Bandwidth

100 MHz

200 MHzTwo channel series have a single 1 GSa/s ADC, four channel series have two 1 GSa/s ADCs. When all
Sampling Rate (Max.)

channels are enabled, each channel has a maximum sample rate of 500 MSa/s. When a single channel
per pair is active, that channel has sample rate of 1 GSa/s
4 (four channel series)

Channels

2+EXT (two channel series)
7 Mpts/CH (not interleave mode);

Memory Depth (Max.)

14 Mpts/CH (interleave mode)

Waveform Capture Rate (Max.)

100,000 wfm/s (normal mode), 400,000 wfm/s (sequence mode)

Trigger Type

Edge, Slope, Pulse Width, Window, Runt, Interval, Dropout, Pattern, Video

Serial Trigger and decoder (Standard)

I2C, SPI, UART/RS232, CAN, LIN

16 Digital Channels (four channel series only, optional)

Maximum waveform capture rate up to 1 GSa/s, Record length up to 14 Mpts/CH

USB AWG module (four channel series only, optional)

One channel, 25 MHz, sample rate of 125 MHz, wave length of 16 kpts

Bode plot ( four channel series only)

Minimum start frequency of 10 Hz, minimum scan bandwith of 500 Hz, maximum scan bandwidth of 120
MHz (dependent on Oscilloscope and AWG bandwidth), 500 maximum scan frequency points

I/O

USB Host, USB Device, LAN, Pass/Fail, Trigger Out, Sbus (Siglent MSO)

Probe (Std)4 pcs passive probe PP510

Display

7 inch TFT -LCD (800x480)

Weight4/2 pcs passive probe PP215

Four channel series: Without package 2.6 Kg; With package 3.8 Kg
Two channel series: Without package 2.5 Kg; With package 3.5 Kg

Ordering information
SDS1104X-E 100 MHz Four Channels
Product NameSDS1204X-E 200 MHz Four Channels
SDS1202X-E 200 MHz Two Channels
USB Cable -1
Quick Start -1

Standard Accessories

Passive Probe -2/4
Certification -1
Power Cord -1

Optional Accessories

16 Channels MSO Software (four channel series only)

SDS1000X-E-16LA

16 Channels Logic Analyzer (four channel series only)

SLA1016AWG Software (four channel series only)

SDS1000X-E-FG

USB AWG Module Hardware (four channel series only)

SAG1021I

STB Demo Source

STB-3

High Voltage Probe

HPB4010

Current Probes
Differential Probes

CP4020/CP4050/CP4070/CP4070A/CP5030/CP5030A/
CP5150/CP5500
DPB4080/DPB5150/DPB5150A/DPB5700/DPB5700A

www.siglent.com 55

Digital Oscilloscope

SDS1000DL+/CML+ Series Digital Oscilloscope
Application• Electronic circuit design and debugging• Electrical circuit function test
• Inspect instantaneous signal• Industrial control and measuring• Products quality control• Education and training

Key Features• 50 MHz to 150 MHz Bandwidth• 500 MSa/s~1 GSa/s sampling rate,32 Kpts~2 Mpts memory depth• 7 inch (8*18 div) color TFT-LCD display• 6 digits hardware frequency counter, real time counting display• Waveform record and play back function• Unique digital filter and data recorder function• Embedded 12 languages, online help, one key storing and one key printing• Interface: USB Device, USB Host, LAN, Pass/Fail• Supports USB-TMC protocol and SCPI programming command control

www.siglent.com 56

Digital Oscilloscope
Specifications
Model

SDS1052DL+

SDS1072CML+

SDS1102CML+

SDS1152CML+

Bandwidth

50 MHz70 MHz

100 MHz

150 MHzChannels

2 CH +1 EXT

Real time sampling rate

500 MSa/s

1 GSa/s

1 GSa/s

1 GSa/s

Equivalent sampling rate

50 GSa/s

Memory depth

32 Kpts

2 Mpts2 Mpts2 Mpts

Input impedance

1 MΩ||17 pF

1 MΩ||17 pF

1 MΩ||17 pF

1 MΩ||17 pF

Vertical sensitivity

2 mv~10 v/div

2 mv~10 v/div

2 mv~10 v/div

2 mv~10 v/div

Vertical resolution

8 bit

Trigger source

CH1, CH2, Ext, Ext/5, AC Line

Trigger types

Edge, Pulse, Video, Slope, Alternative

Math operation+, -, *, /, FFT

Digital filter

High pass, Low pass, Band pass, Band stop

Data recorder function

√

√

√

Max input voltage± 400 V (DC+AC Pk-Pk)

Internal storage

2 groups of reference waveform, 20 groups of setting,10 groups of waveform

External storage

Bitmap save, CSV save, Waveform save, Setting save

Lasting

Turn off, 1 s, 2 s, 5 s, infinite

Language

English, French, German, Russian, Spanish, Simplified Chinese, Traditional Chinese, Portuguese, Japanese, Korean, Italian, Arabic

Interface

USB Host, USB Device, LAN, Pass/Fail

Display

7 inch color TFT-LCD

Power

AC 100-240 V, 45 Hz-440 Hz, 50 VA Max√

Normal Memory (40 kpts)

Long Memory (2 Mpts)

32 types of auto measurements

5 parameters display

Zoom Function

Pass/Fail Function

Math Function

Embedded Online Help

Standard Accessories

www.siglent.com 57

Handheld Oscilloscope

SHS1000X/SHS800X
Handheld Oscilloscope
Application• 200 MHz, 100 MHz bandwidth models• Sample rate of 1 GSa/s (single-channel), Sample rate of 500 MSa/s (two-channels)• The Siglent SPO technology• Waveform capture rates up to 100,000 wfm/s (normal mode) and 400,000 wfm/s (sequence mode)• Supports 256-level intensity grading and color temperature display modes• Record length up to 12 Mpts
• Digital trigger system• Intelligent trigger: Edge, Slope, Pulse Width, Window, Runt, Interval, Time out (Dropout), Pattern• Serial bus triggering and decoding (Standard) for IIC, SPI, UART, CAN, and LIN protocols• Video trigger/HDTV• Low background noise with voltage scales from 2 mV/div to 100 V/div• 3 one-button shortcuts for Oscilloscope, Multimeter and Recorder functions• 8 one-button shortcuts for: Run/Stop, Auto Setup, Default, Measure, Cursors, Display/Persist, Clear Sweep and Print. More function shortcuts
available when combined with the shift button• Segmented acquisition (Sequence) mode, divides the maximum record length into multiple segments (up to 80,000), according to trigger
conditions set by the user, with a very small dead time segment to capture the qualifying event• History waveform record (History) function (maximum recorded waveform length is 80,000 frames)• Automatic measurement function for 38 parameters as well as Measurement Statistics, Zoom, Gating, Math, History and Reference functions• 1 Mpts FFT. Support Peaks and Markers• Math and measurement functions use all sampled data points (up to 12 Mpts)• Math functions (FFT, addition, subtraction, multiplication, division, integration, differential, square root)• Default key can be customized for user settings or factory “defaults”• Supports Multi-language display and embedded online help
• Security Erase mode
• Search and navigate function• Includes Recorder mode, including Sample and Measurement Loggers• 6000 counts Digital Multimeter, Support DCV, ACV, DCI, ACI, Resistance, Diode, Capacitance, Continuity test
• True RMS AC Voltage/Current measurement multimeter• 5.6-inch TFT-LCD display with 640 * 480 resolution• Interface types: Isolated USB Host, USB Device (MicroUSB -TMC)• Supports SCPI remote control commands• UL2054 certified lithium battery pack, 6900 mAh capacity, external charger• IP Rating: IP51
• Compliance with UL61010-1, UL61010-2-030, UL61010-2-033
www.siglent.com 58

Handheld Oscilloscope
Characteristics• Front panel and back panel

Front panel of the SHS1000X series

Rear of the SHS800X series

Large bright 5.6-inch TFT -LCD display with 640 * 480 resolution. The most commonly used functions are accessible using 8 different one-button operation keys:
Run/Stop, Auto Setup, Default, Cursor, Measure, Display/Persist, Clear Sweep, and Print. More function shortcuts are available combined with the shift button.

• When two channels are enabled, each channel has a maximum sample rate of 500 MSa/s. When a single
channel is active, that channel has a sample rate of 1 GSa/s• Record Length of up to 12 Mpts

• Waveform Capture Rate up to 400,000 wfm/s

Using hardware-based Zoom technologies and max record length up toWith a waveform capture rate of up to 400,000 wfm/s (sequence mode),

12 Mpts, users can oversample to capture for longer periods at higher

the oscilloscope can easily capture unusual or low-probability events.

resolution and use the zoom feature to see more details within eachsignal.
www.siglent.com 59

Handheld Oscilloscope• 256-Level Intensity Grading and Color Temperature Display

SPO display technology provides fast refresh rates. The resulting

The color temperature display is similar to the intensity-graded trace

intensity-graded trace is brighter for events that occur with morefunction, but the trace occurrence is represented by different colors (color

frequency and dims when the events occur with less frequency.

“temperature”) as opposed to changes in the intensity of one color. Red
colors represent events that occur more frequently, while blue is used tomark points that occur less frequently.

• History Waveforms (History) Mode and• Serial Bus Decoding Function

Segmented Acquisition (Sequence)

SHS800X/SHS1000X displays the decoding through the events list. Bus

Playback the latest triggered events using the history function.

protocol information can be quickly and intuitively displayed in a tabular

Segmented memory collection will store trigger events into multiple

format.

(Up to 80,000) memory segments, each segment will store triggered
waveforms and timestamps for each frame.

• True measurement to 12 M points• Gate and Zoom Measurement

SHS800X/SHS1000X series can measure all sampled data points up to

Through Gate and Zoom measurement, the user can specify an arbitrary

12 Mpts. This ensures the accuracy of measurements while the math co-interval of waveform data analysis and statistics. This helps avoid

processor decreases measurement time and increases ease-of-use.

measurement errors that can be caused by invalid or extraneous data,
greatly enhancing the measurements’ validity and flexibility.

www.siglent.com 60

Handheld Oscilloscope• 1M points used to calculate the FFT• Customizable Default Key

The new math co-processor enables FFT analysis of incoming signals
using up to 1 M samples per waveform. This provides high-frequency
resolution with a fast refresh rate. The FFT function also supports avariety of window functions so that it can adapt to different spectrum

The current parameters of the oscilloscope can be preset to Default Key
through the Save menu.

measurement needs. Support Peaks, Markers, a variety of numbers.

• Search and Navigate• 6000 Counts Digital Multimeter

The SHS800X/SHS1000X series can search events specified by the user

6000 count digital multimeter featured function of DCV, true RMS ACV,

in a frame. It can also navigate by time (delay position) and historical

DCI, ACI, Diode, Resistance, Capacitance, and Continuity.

frames.

• Sample Logger• Measurement Logger

The Sample Logger is the mode of logging the sampling points for a long

The measurement Logger is the mode of logging the measurement value

time. For there are many sampling points to log, they are logged into thefor a long time. For the amount of measurement data is relatively small,internal flash or external U disk in real-time. After stopping logging, theto process quickly, the data is logged in memory. After stopping logging,user can recall the sampling points on the oscilloscope, or analyze thethe data can be saved into the internal flash or external U disk.

saved data on the computer.
www.siglent.com 61

Handheld Oscilloscope• Adapter/Battery

Wall power using the supplied adapter

Battery powered

SHS800X/SHS1000X supports adapter power supply and battery power

SHS800X/SHS1000X uses a UL2054 certified lithium battery package.

supply. After connecting the adapter, the battery enters into charging

The battery capacity of 6900 mAh can guarantee long-term operation

mode. The adapter provides a maximum 4 A output current.

without an external power supply for up-to 5.5 hours (SHS800X) and 4
hours (SHS1000X). The battery supports an external charger to further
meet the requirements of portability.

• Connectivity

Right side of the SHS800X series

Left side of the SHS1000X series

SHS800X/SHS1000X supports USB Host, USB Device (Micro USB -TMC)

Specifications
Model

SHS810X

SHS820X

SHS1102X

SHS1202X

Bandwidth

100 MHz

200 MHz

100 MHz

200 MHz

Sample rate (Max.)
Channels
Memory depth (Max.)

Two-channel share a single 1 GSa/s ADC. When two channels are enabled, each channel has a maximum sample rate of 500 MSa/s.
When a single channel is active, that channel has a sample rate of 1 GSa/s
2 analog oscilloscope channels, 1 multimeter channel
6 Mpts/CH (dual-channel mode)
12 Mpts/CH (single channel mode)

Waveform capture rate (Max.)

100,000 wfm/s (normal mode), 400,000 wfm/s (sequence mode)

Trigger type

Edge, Slope, Pulse Width, Window, Runt, Interval, Dropout, Pattern, Video

Serial Trigger and decoder

IIC, SPI, UART, CAN, LIN

Data Logger(Recorder)
I/O

Max input Voltage(Scope)Sample Logger. The Max sample rate is 25 kSa/s, the Min sample rate is 1 Sa/s
Measurement Logger. The Max interval is 10 minutes, the Min interval is 0.1second. The Max items of logging is 4
USB Host, USB Device
CAT II 300 Vrms Between BNC Signal and Protecting EarthCAT III 600 Vrms, CAT II 1000 Vrms Between BNC Signal and

CAT II 30 Vrms Between BNC GND and Protecting EarthProtecting EarthCAT II 300 Vrms Between BNC Signal and BNC GND

CAT III 600 Vrms, CAT II 1000 Vrms Between BNC GND andProtecting EarthCAT III 300 Vrms Between BNC Signal and BNC GND

Max input Voltage (Meter)

CAT III 300 Vrms, CAT II 600 Vrms

CAT III 600 Vrms, CAT II 1000 Vrms

Probe

PP510

PB925

Display

5.6-inch TFT-LCD (640x480)

WeightWithout package 1.75 kg. With package 3.5 kg

www.siglent.com 62

PP215

Handheld Oscilloscope
Ordering Information
SHS820X 200 MHz
Product NameSHS810X 100 MHz
SHS1202X 200 MHz Isolated Input
SHS1102X 100 MHz Isolated Input
USB Cable -1
Quick Start -1
Passive Probe -2
Multimeter Test Lead -2

Standard Accessories

Certification -1
Power Adapter -1
Battery -1
SCD600MA Current Measurement Adapter -1
SCD10A Current Measurement Adapter -1
Carrying Bag -1

Optional Accessories

STB Demo Source

STB-3

High Voltage Probe

HPB4010

Current Probes

CP4020/CP4050/CP4070/CP4070A/CP5030/CP5030A/CP5150/
CP5500/CPL5100

Differential Probes

DPB1300/DPB4080/DPB5150/DPB5150A/DPB5700/DPB5700A

Smart Battery Charger

GSCH4000A

www.siglent.com 63

Waveform Generator

SDG7000A Arbitrary Waveform Generator

Key Features• Dual channel differential/single-ended output, 16-bit LVDS/LVTTL digital bus output• High-performance sampling system with 5GSa/s sample rate and 14 -bit vertical resolution• 1 GHz maximum bandwidth• Generates arbitrary waveform with sample rates of 0.01 Sa/s ~ 2.5 GSa/s, with maximum memory depth of 512 Mpts, and provides segment editing /playback
functions• Generates vector signals with up to 500 MS/s symbol rate
• Generates low jitter pulses with 1 ns minimum pulse width and 500ps minimum edge
• Up to 1 GHz bandwidth White Gaussian Noise and the bandwidth is adjustable• Supports PRBS up to 312.5 Mbps
• The digital bus can output digital signals up to 1 Gbps
• Supports analog/digital modulation, sweeping and bursting• Enhanced dual channel operation functions: inter channel tracking, coupling and copying; Dual channel superposition function; Supports mutual modulation
between channels• The 24 Vpp analog output is superimposed with ± 12 Vdc offset to provide a maximum output range of ± 24 V (48 V)• High precision Frequency Counter• 5-inch capacitive touch screen with resolution of 800x480; Supports external mouse and keyboard operation; Supports WebServer to control the instruments
remotely• Supports multiple interfaces: 10MHz In, 10MHz Out, Trigger In/Out, Markers etc• Supports SCPI command for easy integration into test systems

www.siglent.com 64

Waveform Generator
Characteristics• Multi-functional Waveform Generator

The SDG7000A series integrates multiple waveform generator functions from DC to continuous waves up to 1 GHz, which can replace RF signal generators in some
applications. It adopts Siglent’s TrueArb point-by-point arbitrary waveform generation technology, which enables user-adjustable output sample rates from 0.01
Sa/s to 2.5 GSa/s with excellent jitter performance and the generation of I/Q vector signals with a maximum settable bandwidth greater than 500 MHz. Using thebenefits of Siglent’s EasyPulse architecture, a low jitter pulse with a minimum pulse width of 1 ns can be generated. The SDG7000A also features a Gaussian noise
output with adjustable bandwidth, Pseudo-random code generation, an optional 16 channels of digital signal output for synthesizing digital communications, andmuch more.

• Wide Range Amplitude Output

24Vpp analog output superimposed with ± 12 Vdc offset, providing a maximum output range of ± 24 V (48 V).

www.siglent.com 65

Waveform Generator• Excellent Arbitrary Waveform Generation
AFG mode
uses traditional DDS technology to generate arbitrary waveforms

AWG mode
uses the innovative TrueArb technology, with an adjustable sample rate from 0.01 Sa/s~ 2.5 GSa/s and jitter less than 20 ps. It not only has all the advantagesof traditional DDS technology, but also overcomes its intrinsic jitter and distortion defects. The flexible platform also provides zero order hold, linear and sincinterpolation methods for increased flexibility when creating complex waveforms.

Sequence editing and playback

EasyWaveX

The SDG7000A supports up to 1024 arbitrary wave segments, eachsupports extensive arbitrary wave editing functions including manual,of which can be set with a maximum of 65535 repetitions. Whenlinear, coordinate, and equation drawing that facilitate rapid generation ofswitching between segments, the output seamlessly moves from thethe required waveforms. The EasyWaveX editing software is embedded inlast point of the previous segment to the first point of the next segment

the SDG7000A, and can also be installed in a computer, interacting withwithout generating an idle level. It is suitable for applications with highthe SDG7000A over USB or LAN interfaces.

requirements for waveform switching.

• High-Speed Low Jitter Pulse

Low jitter
When a Square/Pulse waveform is generated
by traditional DDS, there can be additional
jitter if the sample rate is not an integer-related
multiple of the output frequency. EasyPulse
technology successfully overcomes this weakness
in DDS designs and helps to produce low jitter
Square/Pulse waveforms.

High speed

Flexible edge

The minimum 1 ns pulse width, can be generated at any frequency. The

Adjustable fine step resolution to 100 ps. The minimum edge is 500 ps

pulse width can be finely adjusted in steps of 10 ps.

and can be generated at any frequency. The rising/ falling edge can be
set respectively and can be used to generate asymmetric pulse

www.siglent.com 66

Waveform Generator• Vector Signal Output (Optional)

The SDG7000A can generate common modulation types of IQ signals, such as ASK, FSK, PSK, QAM. With the innovative resampling technology, excellent EVM
performance can be obtained at any symbol rate between 250 S/s ~ 500 MS/s. The built-in digital quadrature modulator can modulate the carrier of the IQ signal toany frequency between 0 Hz~1 GHz. The EasyIQ software can be used to generate and edit various types of IQ signals.

• Complex Signal Generator

Modulation

Sweep and Burst

A variety of analog and digital modulation modes such as AM, FM,

Sweep supports "Line" and "Log" modes, while Burst enables "NCycle"

PM, FSK, ASK, PSK, DSB-SC, and PWM are supported. There are threeand "Gated" modes. Both Sweep and Burst support trigger sources:

modulation sources: Internal, External, and Channel.

Internal, External, and Manual.

Harmonics Function
provides the ability to add higher-order elements to your signal.

www.siglent.com 67

Waveform Generator• 16 Channel Digital Output (Optional)

Purchase the corresponding digital bus kit to get 16-channel LVTTL or LVDS output with a bit rate of 1 μbps ~ 1 Gbps. Combine the digital bus with the analog
channels to realize mixed-signal outputs.

• Enhanced Dual Channel Functionality
Two Dual-Channel Operation Mode
Independent mode
enables the two channels to be used as two independent generators. Independent mode also eliminated the discontinuity on the output
when parameters (frequency, amplitude) change.

Phase-Locked mode
Automatically aligns the phases of each output.

Track/Copy/Coupling
The track, copy and coupling functions between the two channels can quickly transfer
the parameters of one channel to the other according to the requirements, greatly
simplify the operation and meet the requirements of fast and synchronous switching
waveforms.

www.siglent.com 68

Waveform Generator

Waveform Combining
Superimposes CH1 and CH2 waveforms internally and provides the combined
waveform to a user-selected output. It easily combines basic waveforms, random
noise, modulation signals, sweep signals, burst signals, EasyPulse waveforms, and
TrueArb waveforms without external connections or complex editing.

Channel Modulation
One channel can modulate the other without external connections. This featureprovides an easy method for complex modulation waveform creation. The modulating
wave channel can be directly output and compared with the modulated signal.

• SigIQPro Signal Generation Software (Optional)
SigIQPro is a flexible PC-based signal generation software that takes signal generation
to a whole new level, making it easy to generate complex signals that are fullycompliant with Bluetooth, IoT and other communication standards. SIGLENT
instruments and SigIQPro signal generation software integrate simulation, design andtest to easily meet the needs of users at all stages of design, R&D, and production.

Specifications
Model

SDG7102A

Number of channels

2 Differential/Single-ended

Bandwidth

1 GHz

Sample rate

5 GSa/s

Vertical resolution

14-bitArbitrary waveform

0.01 Sa/s ~ 2.5 GSa/s sample rate; 24 pts ~ 512 Mpts/ch memory depth, with segment editing and playback

Vector signal (Optional)
Continuous waveform
Pulse

SDG7052A

SDG7032A

500 MHz

350 MHz

500 MS/s max symbol rate; Carrier DC ~ 1 GHz settable. Includes modulation modes such as ASK, PSK, FSK and QAM.
EasyIQ software provides vector signal creation and editing
Up to 1GHz, supports harmonic generation function
Min pulse width 1 ns, min. edge 500 ps pulse with low jitter, the rise/fall edge is independently fine adjustable, and thepulse width is fine adjustable

Noise

Bandwidth 1 mHz ~ 1 GHz adjustable

PRBS

Bit rate 1 μbps ~ 312.5 Mbps, length PRBS3 ~ PRBS32

Complex signal generation

Supports internal/external modulation, AM, FM, PM, PWM, FSK, PSK, ASK, etc.; Supports sweep; Support burst

Dual-channel function

Inter channel tracking, coupling, and copying. Dual channel superposition function. Supports mutual modulation between
channels

Output range
Digital bus(Optional)

24 Vpp analog output superimposed ± 12 V DC offset, supports a maximum output range of ± 24 V (48 V)
16-bit, LVTTL or LVDS output
Bit rate: 1 μbps ~ 1 Gbps
USB 2.0 Host x3, USB 2.0 Device(USBTMC)Interface

LAN 10M/100M (VXI-11/Telnet/Socket/WebServer)
EXT MOD/CNT, 10MHz In, 10MHz Out, Marker x2, Trigger In/Out
5” TFT-LCD with capacitive touch screen (800x480)

Interaction

Supports mouse operation
Supports Webserver
Supports SCPI control
www.siglent.com 69

Waveform Generator
Ordering Information
Product Description
SDG7102A

1 GHz, 5 GSa/s, 14-bit, 512 Mpts, 5-inch capacitive touch screen

SDG7052A

500 MHz, 5 GSa/s, 14-bit, 512 Mpts, 5-inch capacitive touch screen

SDG7032A

350 MHz, 5 GSa/s, 14-bit, 512 Mpts, 5-inch capacitive touch screenStandard Configurations
USB cable×1
BNCcoaxial cable×2
Quick start ×1
Power cord ×1
Wireless mouse×1
Optional Configurations

Model

20 dB Attenuator

ATT-20dB

Single Instrument Rack Mount Kit

SSG-RMK

USB-GPIB Adapter

USB-GPIBHigh precision OCXO (Installed at the factory,
cannot be added after purchase)

10M_OCXO_L

Digital Bus Kit-LVTTL

DIG-LVTTL

Digital Bus Kit-LVDS (Without RF cables)

DIG-LVDS

Digital Bus Kit-LVDS (With 32 RF cables)

DIG-LVDS-2

IQ Signal Generator Function (software)

SDG-7000A-IQ

350 MHz to 500 MHz bandwidth upgrade (software)

SDG-7000A-BW05

500 MHz to 1 GHz bandwidth upgrade (software)

SDG-7000A-BW10

www.siglent.com 70

Waveform Generator

SDG6000X Series Pulse/Arbitrary Waveform Generator

Key Features• Dual-Channel, 500 MHz maximum bandwidth, 20 Vpp maximum output amplitude, high fidelity output with 80 dB dynamic range• High-performance sampling system with 2.4 GSa/s sampling rate and 16-bit vertical resolution• Innovative TrueArb technology, based on a point-by-point architecture, supports any 8 pts~8 Mpts Arb waveform with a sampling rate in rangeof 1 μSa/s~75 MSa/s• Innovative EasyPulse technology, capable of generating lower jitter Square or Pulse waveforms, brings a wide range and extremely high precision in pulse
width and rise/fall times adjustment• Multi-function signal generator, meeting requirements in wide range, Continuous Wave Generator, Pulse Generator, Function
Arbitrary Waveform Generator, IQ Signal Generator (optional), Noise Generator, PRBS Generator• Sweep and Burst function• Harmonics function• Waveform Combining function• Channel Coupling, Copy and Tracking function• 196 built-in arbitrary waveforms• High precision Frequency Counter• Standard interfaces include: USB Host, USB Device （USBTMC）, LAN（VXI-11, Socket, Telnet）, GPIB (Optional)• 4.3” touch screen display for easier operationwww.siglent.com 71

Waveform Generator
Characteristics• Continuous Wave

Up to 500 MHz continuous sine wave.

• PulseAdjustable Pulse Width
The pulse width can be fine-tuned to the minimum of 3.3 ns with an
adjustment step as small as 100 ps, at any frequency.

Adjustable Edge
The rise/fall times can be set independently to the minimum of 1 ns
at any frequency with a minimum adjustment step as small as 100
ps.

Low Jitter
When a Square/Pulse waveform is generated by traditional DDS,
there can be additional jitter if the sampling rate is not an integerrelated multiple of the output frequency. EasyPulse technology
successfully overcomes this weakness in DDS designs and helps to
produce low jitter Square/Pulse waveforms.
www.siglent.com 72

Waveform Generator• Arbitrary Waveform
Traditional DDS designs can lead to additional jitter and distortion when sourcing arbitrary waveforms. The SIGLENT TrueArb design minimizes jitter and distortion to
help deliver high fidelity arbitrary waveforms.

Point by Point Output

Low Jitter

TrueArb generates arbitrary waveforms point-by-point. It never skips

As with EasyPulse, TrueArb effectively overcomes the clock jitter that can

any point so that it can reconstruct all the details of the waveform, as

effect traditional DDS generators.

defined. Two interpolation modes are available: linear and zero-orderhold.

• PRBS

PRBS3 ~ PRBS32 with finely adjustable 10-6 bps ~ 300 Mbps bit rate and1 ns ~ 1us edge.

Preset common logic levels such as TTL, LVCMOS, LVPECL and LVDS. An
added differential mode provides an easy way to generate differential
signals using the both channels.

www.siglent.com 73

Waveform Generator• IQ (optional)

The SDG6000X supports popular modulation types such as ASK, FSK, PSK, and QAM. Proprietary resampling technology provides excellent EVM performance at
arbitrary symbol rates between 250 Symb/s ~ 37.5 MSymb/s. Built-in digital quadrature modulator provides the possibility to generate IQ signals from baseband to
500 MHz intermediate frequency.

IQ waveforms can be generated by the PC software EasyIQ.

www.siglent.com 74

Waveform Generator• Complex Signals Generation

Modulation

Sweep and Burst

Plenty of modulation types, such as AM, FM, PM, FSK, ASK, PSK, DSB-

Sweep modes include “Linear” and “Log”. Burst modes includes “N

AM, PWM are supported. The modulation source can be configured as

cycle” and “Gated”. Both Sweep and Burst can be triggered by “Internal”,“Internal” or “External”.

“External” or “Manual” source.

Waveform Combining
The waveform combining function
superimposes CH1 and CH2 waveforms
internally and provides the combined
waveform to a user-selected output. Easily
combine basic waveforms, random noise,
modulation signals, sweep signals, burst
signals, EasyPulse waveforms and TrueArb
waveforms.

Harmonics Function
Harmonics function gives you the ability to add
higher-order elements to your signal.

• Two Dual-channel Operation Mode

• Frequency Counter"Phase-Locked" mode automatically aligns the phases of each output.

8-digit hardware frequency counter with statistics function and input

While "Independent" mode permit the two channels to be used as two

range of 0.1 Hz ~ 400 MHz.

independent generators. Independent mode also smoothes parameter(frequency, amplitude) changes made to an active channel.
www.siglent.com 75

Waveform Generator
Specifications
Model

SDG6022X

SDG6032X

SDG6052X

Bandwidth

200 MHz

350 MHz

500 MHz

Number of channels

2

Sampling rate

2.4 GSa/s (2X Interpolation)

Vertical resolution

16 bitArbitrary waveform length

2 ~ 20 Mpts

Display

4.3” touch screen display, 480 x 272 x RGB

Interface

Standard: USB Host, USB Device, LAN
Optional: GPIB (USB-GPIB adaptor)

Frequency
Resolution±1 ppm (25℃ )±2 ppm (0-40℃ )1st-year aging±1 ppm (25℃ )10-year aging±3.5 ppm (25℃ )Sine
0~1 MHz ( included ) < -65 dBc
1~60 MHz ( included )< -60 dBc
Harmonic distortion

60~100 MHz ( included ) < -50 dBc
100~200 MHz ( included ) < -40 dBc
200~300 MHz ( included ) < -30 dBc
300 MHz ( included )< -28 dBc

Total Harmonic Distortion
Non-harmonic spurious

10 Hz ~ 20 kHz < 0.075%≤350 MHz < -60 dBc>350 MHz < -55 dBc

Pulse
Frequency

1 μHz ~ 150 MHz (SDG6052X, SDG6032X)
1 μHz ~ 80 MHz (SDG6022X)

Pulse Width≥3.3 ns

Pulse width accuracy±(0.01%+0.3 ns)

Rise time ( setting range )1 ns (10% ~ 90%) SDG6052X, SDG6032X
2 ns (10% ~ 90%) SDG6022X

Overshoot

3%,100 kHz, 1 Vpp, 50 Ω load , 2 ns edge

Duty cycle

0.001% ~ 99.999% Limited by frequency setting

Duty cycle resolution

0.001%

Jitter (rms) cycle to cycle<100 ps, 1 Vpp, 50 Ω load

Arbitrary Wave
Frequency setting range

1 μHz ~ 50 MHz

Waveform length

2 pts ~ 20 Mpts

Sampling rate

1 uSa/s ~ 300 MSa/s (TrueArb mode)
1.2 GSa/s (DDS mode)

Vertical resolution

16 bitJitter (rms) cycle to cycle≤100 ps (1 Vpp, 50 Ω load , TrueArb mode)www.siglent.com 76

Waveform Generator
SquareFrequency

1 μHz~ 120 MHz (SDG6052X, SDG6032X)
1 μHz~ 80 MHz (SDG6022X)

Rise /fall times

2 ns~2.4 ns (10% ~ 90%, 1 Vpp, 50 Ω load)

Overshoot≤3% (100 kHz, 1 Vpp, 50 Ω load)

Duty cycle

10% ~ 90% (Limited by frequency setting)

Jitter (rms) cycle to cycle<100 ps (1 Vpp, 50 Ω load)

OutputAccuracy±(1%+1 mVpp) (10 kHz sine, 0 V offset)

Amplitude flatness±0.3 dB (50 Ω load, 0.5 Vpp, compare to 1 MHz Sine)

Output impedance

50±0.5 Ω (100 kHz sine)

Output current-200 ~ 200 mA

Crosstalk< -60 dBc (CH1=CH2=0 dBm, Sine, 50 Ω load)

IQ (optional)
Symbol rate

250 Symb/s ~ 37.5 MSymb/s (Limited by the oversampling factor)

Vertical resolution

16 bitModulation type2ASK, 4ASK, 8ASK, BPSK, QPSK, 8PSK, DBPSK, DQPSK, D8PSK, 8QAM, 16QAM, 32QAM, 64QAM, 128QAM, 256QAM, 2FSK, 4FSK,
8FSK, 16FSK, MSK, MultiTone, custom (Supported by EasyIQ software)

Pattern

PN7, PN9, PN15, PN23, User file, Custom (Supported by EasyIQ software）

Output Range

1 mVrms ~ 0.5 Vrms (Carrier frequency

500 MHz (IF Output), 50 Ω load)

PRBS
Bit rateSequence length
Rise/fall times

1 ubps~ 300 Mbps (SDG6052X, SDG6032X)
1 ubps~ 160 Mbps (SDG6022X)
2m-1, m = 3, 4, ... , 32
1 ns ~ 1 us (SDG6052X, SDG6032X. 10% ~ 90%, 1 Vpp, 50 Ω load)
2 ns ~ 1 us (SDG6022X. 10% ~ 90%, 1 Vpp, 50 Ω load)
2 mVpp ~ 20 Vpp≤(40 Mbps, HiZ load)

Output Range (Note)

2 mVpp ~ 10 Vpp (40 ~ 240 Mbps ( included ), HiZ load)
2 mVpp ~ 5 Vpp (240 Mbps, HiZ load)

Ordering Information
Product Description
SDG6052X

500 MHz, 2-CH, 2.4 GSa/s, 16-bit

SDG6032X

350 MHz, 2-CH, 2.4 GSa/s, 16-bit

SDG6022X

200 MHz, 2-CH, 2.4 GSa/s, 16-bit

Standard Configurations
Quick start ×1
Power cord ×1
Calibration certificate ×1
USB cable ×1
BNC coaxial cable x2
Optional Configurations
SPA1010

10 W Power Amplifier

ATT-20dB

20 dB Attenuator

USB-GPIB

USB-GPIB Adapter

SDG-6000X-IQ

IQ Signal Generator Functionwww.siglent.com 77

Waveform Generator

SDG2000X Series Function/Arbitrary
Waveform Generator

Key Features• Dual-channel, 120 MHz maximum bandwidth, 20 Vpp maximum output amplitude, high fidelity output with 80 dB dynamic range• High-performance sampling system with 1.2 GSa/s sampling rate and 16-bit vertical resolution. No detail in the waveforms will be lost
• Innovative TrueArb technology, based on a point-by-point architecture, supports any 8 pts~8 Mpts Arb waveform with a sampling rate in rangeof 1 μSa/s~75 MSa/s• Innovative EasyPulse technology, capable of generating lower jitter Square or Pulse waveforms, brings a wide range and extremely highprecision in pulse width and rise/fall times adjustment• Plenty of analog and digital modulation types: AM、DSB-AM、FM、PM、PSK、FSK、ASK and PWM• Practical functions: Channel Copy, Channel Coupling, Channel Track, harmonic generator, overvoltage protection function• Sweep and Burst function, Harmonics mode supported• High precision Frequency Counter• Standard interfaces: USB Host, USB Device（USBTMC）, LAN（VXI-11）• Optional interface: USB-GPIB• 4.3” touch screen display for easier operationwww.siglent.com 78

Waveform Generator
Characteristics• Excellent Analog Channel Performance输出白噪声测试获得的系统-3dB带宽在120MHz以上

The bandwidth of analog channels proves to be greater than
120 MHz, via doing a frequency response test with white noise.

High fidelity sine output. Almost no spurious observed@60 MHz, 0 dBm.

Capacity of outputting large signal at high frequency. Dual-channel,
20 Vpp amplitude can be guaranteed even @20 MHz.

Low noise floor, improves signal-noise ratio.

www.siglent.com 79

Waveform Generator• Innovative EasyPulse Technology

1.001 MHz Pulse

1.001 MHz Pulse

Period Jitter= 64.6 ps rms

When a Square/Pulse waveform is generated by DDS, there will be a one-clock-jitter if the sampling rate is not an integer-related multiple of the output frequency.
SDG2000X EasyPulse technology successfully overcomes this weakness in DDS designs and helps to produce low jitter Square/Pulse waveforms.

The Pulse width can be fine-tuned to the minimum of 16.3 ns

The rise/fall times can be set independently to the minimum of 8.4 ns atwith the adjustment step as small as 100 ps.

any frequency and to the maximum of 22.4 s. The adjustment step is as• Innovative TrueArb Technology

small as 100 ps.

For arbitrary waveforms, TrueArb not only has all the advantages of traditional DDS, but also eliminates the probability that DDS may cause serious jitter anddistortion.

TrueArb generates arbitrary waveforms point by point, never skips any

As with EasyPulse, TrueArb effectively overcomes the defect that DDS

point so that it can reconstruct all the details of the waveform as defined.

may cause the one-clock-jitter in arbitrary waveforms.

• 4.3” Touch Screen Display• Arbitrary Waveform Software EasyWave

4.3” touch screen display, makes operation much more convenient.

EasyWave is a powerful arbitrary waveform editing software that
supports several ways to generate arbitrary waveform such as manual
drawing, line-drawing, equation-drawing, coordinate-drawing, etc. It isquite convenient for users to edit their own arbitrary waveforms through
EasyWave.

www.siglent.com 80

Waveform Generator
Characteristics• High-performance Sampling System
Benefiting from a 1.2 GSa/s and 16-bit sampling system, SDG2000X achieves extremely high accuracy performance in both time domain and amplitude, which
results in more accurately reconstructed waveforms and lower distortion.

1.2 GSa/s Sampling Rate, 16-bit Resolution-SDG2000X
300 MSa/s Sampling Rate, 14-bit Resolution

Specifications
Product Model

SDG2042X

SDG2082X

SDG2122X

Bandwidth

40 MHz80 MHz120 MHz

Sampling rate

1.2 GSa/s (4 X Interpolation)

Vertical resolution

16 bitNum. of channels

2Max. amplitude±10 V

Display

4.3” touch screen display, 480 x 272 x RGB

Interface

Standard: USB Host, USB Device, LAN
Optional: GPIB (USB-GPIB adaptor)

Frequency Characteristics
Parameter

Min.

Typ.

Resolution
Initial accuracy

-1

Max.

Unit

1μ

Hz

+1

ppm

Condition
25℃

-2

+2

ppm

0~40℃

1st-year aging

-1

+1

ppm

25℃

10-year aging

-3.5

+3.5

ppm

25℃

Max.

Unit

Condition

120 M

Hz

-65

dBc

0 dBm, 0~10 MHz (Included)

-60

dBc

0 dBm, 10~20 MHz (Included)

-55

dBc

0 dBm, 20~40 MHz (Included)

-50

dBc

0 dBm, 40~60 MHz (Included)

-45

dBc

0 dBm, 60~80 MHz (Included)

-40

dBc

0 dBm, 80~100 MHz (Included)

-38

dBc

0 dBm, 100~120 MHz (Included)

0.075

%

0 dBm, 10 Hz ~ 20 kHz

-70

dBc

≤50 MHz-65

dBc>50 MHz

Sine Characteristics
Parameter

Min.

Frequency

1μHarmonic distortion

Total Harmonic Distortion
Non-harmonic spuriousTyp.

www.siglent.com 81

Waveform Generator
Square Characteristics
Parameter

Min.

Frequency

1μ

Max.

Unit

25 M

Hz

Rise/fall times

9ns

10% ~ 90%, 1 Vpp, 50 Ω Load

Overshoot3%

100 kHz, 1 Vpp, 50 Ω Load

99.999%Limited by frequency setting

150

ps

1 Vpp, 50 Ω Load

Max.

Unit

Condition

25 M

Hz

Duty cycle

Typ.

0.001

Jitter (rms), Cycle to cycle

Condition

Pulse Characteristics
Parameter

Min.

Frequency

1μPulse width

16.3

Typ.

ns

Pulse width accuracy
Rise/fall times±(0.01%+0.3 ns)
8.4 n

OvershootDuty cycle

0.001

Duty cycle resolution

0.001

10% ~ 90%, 1 Vpp, 50 Ω Load, Subject to pulse

22.4

s

3

%

100 kHz, 1 Vpp

99.999%Limited by frequency setting

width limits%

Jitter (rms) cycle to cycle

150

ps

1 Vpp, 50 Ω Load

Condition

Arbitrary Wave characteristics
Parameter

Min.

Max.

Unit

Frequency

1μ

20 M

Hz

Waveform length

8

8M

pts

1μ

75 M

Sampling rate
Vertical solution

Typ.

Sa/s

TrueArb mode

300

MSa/s

DDS mode

16

bitjitter (rms)

150

ps

1 Vpp, 50 Ω Load, TrueArb mode

Max.

Unit

Condition

Output Characterisics
Parameter

Min.

Typ.

Range

2m

20

Vpp

≤20 MHz, HiZ load(Note 1)

2m

10

Vpp

>20 MHz, HiZ load

1m

10

vpp

≤20 MHz, 50 Ω load

1m

5

vpp

Accuracy± (1%+1 mVpp)

Amplitude flatness

-0.3

+0.3

dB

-0.4

+0.4

dB

50.5

Ω

200

mA

-60

dBc

Output impedance

49.5

Output current-200>20 MHz, 50 Ω load
10 kHz sine, 0 V offset

50

Crosstalk

0~100 MHz (Included), 50 Ω load, 2.5 Vpp,
compare to 10 kHz Sine
100~120 MHz (Included), 50 Ω load, 2.5 Vpp,
compare to 10 kHz Sine
10 kHz sine
CH1 - CH2/CH2 - CH1

Note 1: The specification will be divided by 2 while applied to a 50 Ω load.

Ordering Information
Product Description
Product code

SDG2000X Series Function/Arbitrary Waveform Generator
SDG2042X

40 MHz

SDG2082X

80 MHzSDG2122X

120 MHz

Standard configurations

A Quick Start, A Power Cord, A USB Cable, A Calibration Certificate, A BNC Coaxial Cable

Optional configurations

USB-GPIB adapter

www.siglent.com 82

Waveform Generator

SDG1000X
Function/Arbitrary Waveform Generator

Application• IC test• Simulate sensor• Simulate environment signals• Electrical circuit function test
• Education and training

Key Features• Dual-channel, with bandwidth up to 60 MHz, and amplitude up to 20 Vpp• 150 MSa/s sampling rate, 14-bit vertical resolution, and 16 kpts waveform length• Innovative EasyPulse technology, capable of generating lowerjitter Pulse waveforms, brings a wide range and extremely high precision inpulse width and rise/fall times adjustment• Innovative TrueArb technology, based on a point-by-point architecture, supports any 8 pts~8 Mpts Arb waveform with a sampling rate inrange of 1 μSa/s~75 MSa/s• Special circuit for Square wave function, can generate Square waves up to 60 MHz with jitter less than 300 ps+0.05 ppm of period• Plenty of analog and digital modulation types: AM, DSB-AM, FM, PM, FSK, ASK, PSK and PWM
Sweep and Burst functions• Harmonics Generator function• Waveform Combining function• High precision Frequency Counter• Standard interfaces: USB Host, USB Device (USBTMC), LAN (VXI-11)• Optional interface: GPIB
• 4.3” TFT-LCD display

www.siglent.com 83

Waveform Generator
Models and Key Specifications
Product Model

SDG1032X

SDG1062X

Bandwidth

30 MHz60 MHzSampling rate

150 MSa/s

Vertical resolution

14-bitWaveform Length

16 kptsNum. of channels

2Max. amplitude±10 V

Display

4.3” display, 480 x 272 x RGB

Interface

Standard: USB Host, USB Device, LAN
Optional: GPIB (USB-GPIB adaptor)

Characteristics• Identical dual output-channels with high performance
Capable of outputting large signals at high frequencies. dual-channels, 20 Vpp amplitude can be guaranteed at up to 10 MHz.

• Low Distortion Output
With 0 dBm output, the THD (Total Harmonic
Distortion) is less than 0.075%. Harmonics andspurs are less than -40 dBc throughout the entire
bandwidth.

www.siglent.com 84

Waveform Generator• Innovative EasyPulse Technology

When a Pulse waveform is generated by a common DDS generator, there will be a one-clock-jitter if the sampling rate is not an integer-related multiple of the output
frequency. SDG1000X EasyPulse technology successfully overcomes this weakness in DDS designs and helps to produce low jitter Pulse waveforms.

The rise/fall times can be set independently to the minimum of 16.8 ns at any frequency and to the maximum of 22.4 s. The adjustment step is as small as 100 ps.
The Pulse width can be fine-tuned to the minimum of 32.6 ns with the adjustment step as small as 100 ps.

• High performance Square Waves

Benefitting from a special square-wave generating circuitry, the Square from the SDG1000X breaks the 60 MHz bandwidth barrier, reaching rise/fall times of less than
4.2 ns, and frequencies up to 60 MHz.

The Square wave exhibits the same excellent jitter
performance as the Pulse waveform.

www.siglent.com 85

Waveform Generator
Characteristics• Modulation• Frequency Counter

Multiple modulation types: AM, DSB-AM, FM, PM, FSK, ASK, PSK andHigh precision Frequency Counter with an input frequency range of

PWM. The modulation source can be configured as “Internal” or “External”.

0.1 Hz~200 MHz.

• Harmonics Function• Waveform Combining

Up to 10 harmonics may be generated. Amplitude and phase of each

Capable of combining the waveforms of 2 channels from internal,

harmonic can be set independently.

providing more flexible tools to generate complex waveforms.

Ordering Information
Product Description
30 MHz, 2 CH, 150 MSa/s, 14 bit

SDG1032X

60 MHz, 2 CH, 150 MSa/s, 14 bit

SDG1062X

Standard configurations
Quick Start -1
Power Cord-1
Calibration Certificate -1
USB Cable -1
Optional configurations
BNC Coaxial Cable

SDG-BNC

20 dB Attenuator

ATT-20dB

USB-GPIB Adapter

USB-GPIB

www.siglent.com 86

Waveform Generator

SDG800Function/Arbitrary Waveform Generator
Application• Simulate sensor• Simulate environmental signal• Circuit function test
• IC chip test
• Research and education

Key Features• Advanced DDS technology,125 MSa/s sampling rate, 14 bit vertical resolution• Single channel output, 5 kinds of standard waveforms, built-in 46 kinds of arbitrary waveforms (including DC)• Complete modulation functions: AM, DSB-AM, FM, PM, FSK, ASK, PWM, linear/logarithmic sweep and burst• Innovative EasyPulse technology, can output pulse of low jitter, quick rising/falling edge
• Standard interfaces: USB Device, USB Host, support U-Disk storage and software update• Provide 10 nonvolatile storage spaces for user’s arbitrary waveforms• Be capable of seamlessly connected to SIGLENT Digital Storage Oscilloscope• Configurable with powerful arbitrary waveform editing software EasyWave

www.siglent.com 87

Waveform Generator
Specifications
Model

SDG805

SDG810

SDG830Maximum output frequency

5 MHz

10 MHz30 MHzOutput channels

1

Sampling rate

125 MSa/s

Wave length

16 kpts

Frequency resolution

1 μHz

Vertical resolution

14 bitWaveform

Sine, Square, Ramp, Pulse, Gaussian white noise, Arbitrary waveform, 46 types of built-in arbitrary waveforms

Sine wave

1 μHz ~ 5 MHz

1 μHz ~ 10 MHz

1 μHz ~30 MHz

Square wave

1 μHz ~ 5 MHz

1 μHz ~ 10 MHz

1 μHz ~10 MHz

Pulse

500 μHz ~ 5 MHz

500 μHz ~ 5 MHz

500 μHz ~5 MHz

Ramp/Triangular

1 μHz ~ 300 kHz

1 μHz ~ 300 kHz

1 μHz ~ 300 kHz

Gaussian white noise>5 MHz bandwidth (-3 dB)>10 MHz bandwidth (-3 dB)>30 MHz bandwidth (-3 dB)

Arbitrary waveform

1 μHz ~ 5 MHz

1 μHz ~ 5 MHz

1 μHz ~ 5 MHz

Modulation function

AM, FM, PM, DSB-AM, FSK, ASK, PWM, Sweep, Burst

Standard configuration

USB Host & USB Device

Amplitude (high impedance)www.siglent.com 88

4 mVpp~20 Vpp (≤10 MHz)
4 mVpp~10 Vpp (>10 MHz)DC Power Supply

SPS5000X
Programmable Switching DC Power Supply

Main Features• Rated Output Voltage: 40 V, 50 V, 80 V, 160 V• Rated Output Power: 180 W, 360 W, 720 W, 1080 W• Wide range of output voltage and current, high efficiency power supply• CV, CC priority mode selection, better protection of equipment under test
• Load transient recovery time (Load change from 50~100%) <1 ms• Adjustable slew rate of output voltage and current• Setting and readback resolution: 1 mV, 1 mA
• User enabled internal output discharge circuit to accelerate the down programming of the output voltage• Remote Voltage Sensing• List function up to 50 steps; can be created from the front panel or by importing list sequence files from a USB memory device• External analog voltage and resistor control of voltage or current output• External voltage and current monitoring output• OVP, OCP, LPP, OTP protection.
• 2.4-inch OLED high brightness liquid crystal display, 170-degree viewing angle• Standard Interface: USB, LAN, Analog Control Interface• Optional Interface: USB-GPIB module• 1/2, 1/3, 1/6 rack mount size
• Embedded Web Server offers remote control through a web browser without the need for the driver or software

www.siglent.com 89

DC Power Supply
Design Features• Constant Output Power
In constant output power mode, the voltage and current range is switched automatically to maximize the voltage and current without sacrificing the supply’s output
power. This mode enables the supply to provide a higher output voltage at lower current and a higher output current at lower voltage. Compared to the traditional
rectangular output range of most supplies, the SPS5000X series power supply provides a wider voltage and current output range, which greatly increases theutilization of the power supply.

80V 15A/ 40V 30A Output Operating Area

160V 7.5A Output Operating Area

• Adjustable Output Voltage, Current up/down Slew Rate
The SPS5000X series supports custom setting of the rise/fall slew rate of voltage/current to verify the performance of the object under test as the voltage/current
changes. This feature can effectively prevent the damage caused by inrush current to the DUT in applications such as the testing of capacitive current absorbing
devices.

Output voltage, current up/down slew rate

• CV/CC Priority Mode
When the SPS5000X series power supply is set to CC priority mode, at the power output-on stage, it is able to operate under CC priority to limit the inrush current
spike and overshoot voltage effectively when the power output is turned on.
In CV priority mode, the output voltage reaches the set voltage value quickly. In some applications, such as LED testing, when the power output is started, the surge
current and overshoot voltage will appear when the voltage reaches the on-state voltage of the LEDs.

CV priority modewww.siglent.com 90

CC priority modeDC Power Supply• Intuitive List Operation Function
By editing the single-step setting value, duration, and slew rate, the List function can generate multiple complex sequences to meet complex test requirements. The
user can edit the sequence by 50 steps natively or import the List sequence file via USB for multi-step running.
The minimum precision of delay time is 1ms. The minimum running time is 1 second.

List mode• Rich Interface
The power supply includes USB and Ethernet communication interfaces as standard, and a
USB-GPIB converter module as optional. The embedded Web Server enables control and monitor of the power supply directly from a web browser, eliminating theneed to install software drivers or applications.

Web Server Interface

Specifications
Model

SPS5041X

Output channel

SPS5042X

SPS5043X

1

Rated output voltage

SPS5045X

units

2

3

CH

40

Rated output current

30

60

90

Total rated output power

360

720

1080

Power Ratio

SPS5044X

V
30
720

A
1080

W

3.33
www.siglent.com 91

DC Power Supply
Model

SPS5051X

SPS5081X

SPS5082X

SPS5083X

Output channel

1

Rated output voltage

50

Rated output current

10

15

30

45

Total rated output power

180

360

720

1080

Power Ratio

2.77

Model

SPS5161X

Output channel

SPS5084X

SPS5085X

units

2

3

CH

1
80

V
15

A

720

1080

W

SPS5164X

SPS5165X

units

2

3

CH

3.33
SPS5162X

SPS5163X

1

Rated output voltage

160

Rated output current

7.5

15

22.5

Total rated output power

360

720

1080

Power Ratio

V
7.5
720

A
1080

3.33

Ordering Information
Product information
40 V/30 A 360 W

Product No
Single channel programmable Switching DC Power supply

SPS5041X

40 V/60 A 720 W

Single channel programmable Switching DC Power supply

SPS5042X

40 V/90 A 1080 W

Single channel programmable Switching DC Power supply

SPS5043X

40 V/30 A 360 W X2

Dual Channel Programmable Switching DC Power supply

SPS5044X

40 V/30 A 360 W X3

Three Channel Programmable Switching DC Power supply

SPS5045X

50 V/10 A 180 W

Single channel programmable Switching DC Power supply

SPS5051X

80 V/15 A 360 W

Single channel programmable Switching DC Power supply

SPS5081X

80 V/30 A 720 W

Single channel programmable Switching DC Power supply

SPS5082X

80 V/45 A 1080 W

Single channel programmable Switching DC Power supply

SPS5083X

80 V/15 A 360 W X2

Dual Channel Programmable Switching DC Power supply

SPS5084X

80 V/15 A 360 W X3

Three Channel Programmable Switching DC Power supply

SPS5085X

160 V/7.5 A 360 W

Single channel programmable Switching DC Power supply

SPS5161X

160 V/15 A 720 W

Single channel programmable Switching DC Power supply

SPS5162X

160 V/22.5 A 1080 W

Single channel programmable Switching DC Power supply

SPS5163X

160 V/7.5 A 360 W X2

Dual Channel Programmable Switching DC Power supply

SPS5164X

160 V/7.5 A 360 W X3

Three Channel Programmable Switching DC Power supply

SPS5165X

Standard Accessories
USB Cable -1
Quick Start -1
Calibration Certificate -1
Power Cord -1
Output guard -1
Optional Accessories
SPS5000X-SEC

SPS5000X Series cable

SPS5000X-PAC

SPS5000X Parallel cable

SPS5000X-RMK

SPS5000X EIA Standard rack

www.siglent.com 92

WDC Power Supply

SPD3000
Programmable Linear DC Power Supply

Application• R&D lab general purpose testing• Teaching lab experiment• Automotive electronic test
• Production testing and quality assessment inspection

Key Features (SPD3303X/SPD3303X-E)• 3 independent controlled and isolated output, 32 V/3.2 A×2, 2.5 V/3.3 V/5 V/3.2 A×1, total 220 W• Max 5 digits Voltage, 4 digits Current Display, Minimum Resolution: 1 mV/1 mA• Supports panel timing output functions• 4.3 inch true color TFT- LCD 480x272 display• 3 types of output modes: independent, series, parallel• 100 V/120 V/220 V/230 V compatible design to meet the needs of different power grids.
• Intelligent temperature-controlled fan , effectively reducing noise• Clear graphical interface, with the waveform display function• Internal 5 groups of system parameter save/recall, supports data storage space expansion• Provides PC software: Easypower , supports SCPI , LabView driver

Key Features (SPD3303C)• 3 independent high precision output: 32 V/3.2 A×2, 2.5 V/3.3 V/5 V/3.2 A×1, total 220 W• 4 digits voltage and 3 digits current display, min resolution: 10 mV, 10 mA• Three output modes: independent, series and parallel• 100 V/120 V/220 V/230 V compatible design, to meet the need of different power grids• Smart temperature controlled fan, effectively reduce the noise• Save/Recall 5 group system specifications, support data storage expansion• Connected to PC via USB Device, support SCPI command, to meet the control and communication needs
www.siglent.com 93

DC Power Supply
Specifications
Model

SPD3303C

SPD3303X-E

SPD3303X

CH1: DC voltage range: 0-32 V, DC current range: 0-3.2 A
Channels

CH2: DC voltage range: 0-32 V, DC current range: 0-3.2 A
CH3: DC voltage range: 2.5/3.3/5.0 V, DC current range: 0-3.2 A

Max output power

220 W

Resolution

10 mV / 10 mA

Display digits

1 mV / 1 mA

LED display

4.3 inch TFT-LCD display

4.3 inch TFT-LCD display

4 digits voltage

4 digits voltage

5 digits voltage

3 digits current

3 digits current

4 digits current

CV/CH3: ≤1 mVrms (5 Hz~1 MHz)

Ripple noise

CC: ≤3 mArms

Standard interface

USB Device

Dimension

225 mm (W)×136 mm (H)×275 mm (D)Weight7.5 kg (SPD3303C) 8 kg (SPD3303X/X-E)

USB Device, LAN• Panel displays the timing output
Through front panel operation, 5 groups of timing settings and output control can be displayed, which provides users a simple power programming function. Also a
connection can be made with Siglent’s EasyPower PC software providing a full range of communication and control requirements.

Panel timing output

Real time wave display• Save/Recall setting parameters
SPD3000X series programmable power supply can save or recall 5 groups of setting parameter in internal storage, also supports external storage expansion. You caneasily obtain the settings you needed.

Internal Storagewww.siglent.com 94

PC Timer

DC Power Supply

SPD1000X
Programmable Linear DC Power Supply
Main Features• Single path high-precision programmable voltage output：• 16 V/8 A, total power up to 128 W• 30 V/5 A, total power up to 150 W• Stable, reliable, Low ripple and noise：≤ 350 uVrms/3 mVpp; < 2 mArms• Fast transient response time: < 50 μs• 5 digit Voltage, 4 digit Current Display, Minimum Resolution: 1 mV/1 mA• Supports front panel timing output functions• 2.8 inch true color TFT- LCD 240 *320 display• 2 types of output modes：Two-wire output mode, 4-wire compensation output mode，Maximum compensation voltage 1 V• 100/120/220/230 V compatible design to meet the needs of different power grids• Intelligent temperature-controlled fan reduces noise• Clear graphical interface, with the waveform display function• Internal 5 groups of system parameter save/recall• Includes PC software: Easypower, supports SCPI, LabView driverwww.siglent.com 95

DC Power Supply
Design Features• High-resolution and high-precision output
The SPD1000X power supply features a high measurement resolution of 1 mV/1 mA . This ensures accurate output even with very with smallchanges in voltage or current. This is impossible for a low resolution power supply.

• 4-wire SENSE compensation mode function
In the 4-wire SENSE compensation output mode: By using a separate measurement circuit, the supply can more accurately compensate for any
voltage drops due to high resistance connections or long cables. Maximum compensation voltage is 1 V.

• Low ripple and noise• Fast transient response time

• Low voltage overshoot• 0.01% Load Regulation & 0.2% Line Regulation• Panel displays the timing output• Save/Recall setting parameters
SPD1000X programmable power supply can saveor recall 5 groups of setting parameters in internal
storage. You can easily recall the settings you need.

Panel timing
outputwww.siglent.com 96

Real time wave
displayInternal Storage

PC Timer

DC Power Supply
Specifications
All the specifications are guaranteed when the instrument has been working for more than 30 minutes under the specified operating
temperature. Unless otherwise noted, the specifications are applicable to all the channels of the specified model.

Model

SPD1168X

SPD1305X

DC Output(0 ℃ to 40℃ )Output Voltage: 0 to 16 V
Output Current: 0 to 8 A

Output Voltage: 0 to 30 V
Output Current: 0 to 5 A

Display

2.8 inch true color TFT-LCD
5 digit voltage/4 digit current

Resolution

1 mV/1 mA

Program Accuracy(25 ± 5 ℃ )Voltage: ±(0.03% of reading+10 mV)

Program Accuracy(25 ± 5 ℃ )Voltage: ±(0.03% of reading+10 mV)

Temperature Coefficient per ℃(Output Percentage + Offset)

Voltage: ±(0.01% of reading+3 mV)

Constant Voltage
Mode

Constant Current
Mode

Current: ±(0. 3% of reading+10 mA)

Current: ±(0. 3% of reading+10 mA)

Current: ±(0.01% of reading+3 mA)

Load Regulation≤ 0.01% + 2 mV

Ripple & Noise≤ 350 uVrms/3 mVpp (20 Hz to 20 MHz)

Recovery Time

< 50 μs (50% load change, minimum load 0.5 A)

Line Regulation≤ 0.2% + 3 mA

Load Regulation≤ 0.2% + 3 mA

Ripple & Noise≤ 2 mArms

Locking Key

Yes

Memory Save/Recall

5 SetsMax Output Power128 W

Power Source

AC 100 /120/220/230 V ± 10% 50/60 Hz

Standard Configuration Interface

USB Device, LAN

Insulation

Case to Terminal ≥ 20 MΩ (DC 500 V)
Case to AC line ≥ 30 MΩ (DC 500 V)

Operating Environment

Outdoor Usage:
Elevation: ≤2000 m Environment Temperature 0 to 40 ℃ Relative Humidity ≤ 80%
Installation Level: II Pollution Level: 2

Storage Environment

Environment Temperature: -10 to 70 ℃ Relative Humidity ≤ 70%

Dimension

154.6 (W) × 144.5 (H) × 280(D) mm

Weight≈5.5 kg

150 W

www.siglent.com 97

DC Electronic Load

SDL1000X Series
Programmable DC Electronic Load
Main Feature• SDL1020X (Single channel ): DC 150 V/30 A, total power up to 200 W• SDL1030X (Single channel ): DC 150 V/30 A, total power up to 300 W• 4 static modes / Dynamic mode: CC/CV/CR/CP• CC Dynamic mode: Continuous, pulsed, toggled• CC Dynamic mode: 25 kHz, CP Dynamic mode: 12.5 kHz, CV Dynamic mode: 0.5 Hz• Measuring speed of voltage and current: up to 500 kHz• Adjustable current rise time range: 0.001 A/us~2.5 A/us
• Min. readback resolution: 0.1 mV, 0.1 mA• Short-circuit, Battery test, CR-LED mode, and factory test functions• 4-wire SENSE compensation mode function• List function supports editing as many as 100 steps• Program function supports 50 groups of steps• OCP, OVP, OPP, OTP and LRV protection• External analog control• Voltage, Current monitoring via 0-10 V• 3.5 inch TFT-LCD display, capable of displaying multiple parameters and states simultaneously• Built-in RS232/USB/LAN communication interface, USB-GPIB module (optional)• Waveform trend chart and ease-to-use file storage and call functions• Includes PC software: Supports SCPI, LabView driverwww.siglent.com 98

DC Electronic Load
Design Features• Steady state operating mode

• Dynamic test mode up to 25 kHz (CC)The SDL features four operating modes to provide flexible test

The transient test allows switching between two different load values.

capabilities. In CC mode, the electronic load will sink a constant current,

A common application is to test the dynamic characteristics of a DC

regardless of the voltage at its terminals. In CV mode, the electronic loadsource or DUT (Device Under Test). The transient test function enables

will cause a constant voltage to appear at its terminals. In CR mode, thethe load to periodically switch between two set levels (Level A and Level

electronic load will behave as a fixed resistance value. As shown in theB). The highest frequency can be set to 25 kHz in CC mode. The highest

figure, the electronic load will linearly change the current according tofrequency can be set to 12.5 kHz in CP modes.

the input voltage. In CP mode, the electronic load will cause a constant
power to be dissipated in the load.

• Simplify complex sequencing using the list operation function
You can generate complex load sequences quickly using the list operation function. Here, you can edit the setpoints, dwell time, and slew rate for each step in thetest. *Slew rate can only be edited in CC mode.

• 4-wire SENSE compensation mode function
In CC/CV/CR/CW mode, when a load is connected to a power supply, it
will cause a large voltage-drop on the connection lines between tested
instrument and terminals of load. Using remote sense, you can measure
the voltage at the DUTs input terminals, effectively removing theadditional error due to the voltage drop in the connection wires.

www.siglent.com 99

DC Electronic Load
• Program function• Battery discharge function

In program (auto-test) mode, you can generate a sequence of tests

The SDL1000X can also provide insight into battery performance byusing different modes, mode parameters and durations. This function

analyzing the discharge characteristics of the DUT. The SDL features

is useful for automatically executing a set of tests on a device thenthree stop conditions for the discharge test: Voltage, capacity or time.

display whether the tests passed or failed. Test results are easily viewed

The discharge process is immediately terminated if the stop conditions

by pressing the up and down buttons. The load provides 8 nonvolatile

are met. This provides more control over the test termination and an

registers to save auto-test file for recall later. Each file contains 1-50

extra layer of safety during critical tests. Throughout the test process

steps to set up. Auto-test function is especially useful in the designing

the battery voltage, discharge current, discharge time and discharged

battery charging circuitry.

capability is displayed clearly on the LCD panel.

• OCPT/OPPT Mode

• CR-LED Mode

Over-current protection (OCPT) mode prevents drawing too much

The SDL1000X includes a CR-LED mode specifically for LED driver testing.

current from the DUT. After the input voltage reaches the Von point, theBasing on the traditional CR mode, CR-LED mode adds a diode break-DC load will start to draw a current from the source after a delay time.

over voltage setting. When the input voltage is above this set value, theThe current value will increase by a certain step size at regular intervals.

DC load start to work. Thus, it can emulate the actual characteristics ofSimultaneously, the DC load will compare the input voltage to the OCP

an LED.

voltage:If it is lower, then the present current value will be compared tosee if it is in the current range you have set. Within the range, the OCP
test will evaluate Pass or Fail. If it is outside of the set range, the DC load
will to increase drawing current and compare the voltage again.

Overpower-protection (OPPT) mode: When the input voltage has reachedthe Von point, the load will draw power after a delay time. The power
value will increase by a step size at regular intervals. Simultaneously, theDC load will judge whether the input voltage is lower than OPP voltage
you have set, if it is, then the present current value will be compared
to see if it is in the current range you have set. Within the range, the
OPP test will Pass or Fail. If it is outside of the set power, the load willcontinue to increase the power draw within the cut-off current range andcompare OPP voltage with the input.

www.siglent.com 100

DC Electronic Load
• Voltage Rise/Fall speed test

• Voltage threshold function

The electronic load is also equipped to directly measure voltage rise

The SDL1000X can be set to turn on or off if the input voltage is at,and fall times. It can calculate the time from one voltage to anotherabove, or below a set value. By defining these thresholds, you control

without the need for additional measurement instrumentation. With anwhen the load is active. Which minimizes test time and increases safety.

SDL1000X, you can save money and improve efficiency.

• Waveform trend chart function• Save/Recall setting parameters

The electronic load includes a waveform display function and supports

The load allows you to save different types of files to the internal andthe following operations for the waveform: Pause, recording, andexternal memories. You can recall and read them when necessary.

capturing the waveform. You can quickly observe the trends of parameterchanges as they occur throughout the test.

• External analog control• Multiple protection modes

The load allows the user to control current or voltage through external

The SDL1000X series Programmable DC Electronic Load provides fiveanalog terminals (EXT PRG). Input a 0-10 V analog to adjust 0-100%

protection types: OVP, OCP, OPP, OTP and LRV. When OVP/OCP/OPP/

rated voltage and current. It is very useful for those applications that

OTP/reverse voltage protection (LRV) occurs, the load will immediately

need to change the input value with external signals.

turn off the input and stop sinking. Then, a prompt message is displayed.

www.siglent.com 101

Digital Multimeter

SDM3065X Digital Multimeter

Application• Research Laboratory• Development Laboratory• Detection and Maintenance• Calibration Laboratory• Automatic Production Test

Main Feature (SDM3065X/SDM3065X-SC)• 4.3” TFT-LCD, 480*272• Real 6½ digits readings resolution (2,200,000 counts)• 1Gb Nand flash size, Mass storage configuration files and data files• True-RMS AC Voltage and AC Current measuring• Supports double display, Chinese and English Menu
• File management (support for U-disc and local storage)• Built-in cold terminal compensation for thermocouple• Comes with easy, converient and flexble any sensor measurement control software: EasyDMM• Standard interfaces: USB Device, USB Host, LAN (Optional Accessories: USB- GPIB Adapter )• Scanner Card SC1016 (Only for SDM3065X-SC)• Built-in Hlep system makes information acquisition easier• Support remote control operation via SCPI commands. Compatible with commands of other main stream multimeters• Supports intelligent management system for laboratory based on BS framework and LAN

www.siglent.com 102

Digital Multimeter
Special Features• Histogram• Trend Chart• “Analog” Bar Display• Statistics• Dual Measurement Display• Hold MeasurementdBm Hold Measurement• Interfacewww.siglent.com 103

Digital Multimeter
Scanner card SC1016 (Only for SDM3065X-SC)The SIGLENT Scanner Card SC1016 is a multiplexer that provides multi-point measurement capabilities to the SDM3065X-SC. The scannerfeatures 12 multi-purpose + 4 current channels and supports the following measurement functions: DCV, ACV, DCI, ACI, 2WR, 4WR, CAP, FREQ,
DIODE, CONT and TEMP (RTD and Thermocouple).It provides a convenient and versatile solution for test applications that require multiple
measurement points or signals and is an ideal tool for R&D burn-in and production testing.

Ordering Information
Standard Accessories
Power Cord -1
USB Cable -1
Quick Start -1
warranty Card -1
EasyDMM[1]softwareTest Leads and Alligator Clips -2
Optional Accessories
USB-GPIB

USB-GPIB adapter[1]The latest version of EasyDMM can be downloaded for free from the SDM3000 series of digital multimeter. Please see our web site at www.
siglent.com for more information.

www.siglent.com 104

Digital Multimeter

SDM3055 Digital Multimeter

Application• Research & Development Laboratory• Detection and Maintenance• Calibration Laboratory• Automatic Production Test

Main Features (SDM3055/SDM3055-SC)• Real 5½ digits readings resolution ( 240, 000 counts )• Up to 150 rdgs/s measurement speed• True-RMS AC Voltage and AC Current measuring• 1 Gb Nand flash size, Mass storage configuration files and data files• Built-in cold terminal compensation for thermocouple temperature measurements• With easy, convenient and flexible PC software: EasyDMM• standard interfaces: USB Host, LAN ( Optional Accessories USB-GPIB Adapter )• Scanner Card SC1016 ( Only for SDM3055-SC )• Support remote control operation via SCPI commands.Compatible with commands of main stream multimeters

www.siglent.com 105

Digital Multimeter
Special Features• Histogram• Trend Chart• Bar Chart• Statistics• Dual Display• Hold Measurement• dBm Hold Measurement• Interfacewww.siglent.com 106

Digital Multimeter
Scanner card SC1016 ( Only for SDM3055-SC )The SIGLENT Scanner Card SC1016 is a multiplexer that provides multi-point measurement capabilities to the SDM3055-SC. The scannerfeatures 12 multi-purpose + 4 current channels and supports the following measurement functions: DCV, ACV, DCI, ACI, 2WR, 4WR, CAP, FREQ,
DIODE, CONT and TEMP ( RTD and Thermocouple ). It provides a convenient and versatile solution for test applications that require multiple
measurement points or signals and is an ideal tool for R&D burn-in and production testing.

Ordering Information
Standard Accessories
Power Cord -1
USB Cable -1
Quick Start -1
warranty Card -1
EasyDMM[1]softwareTest Leads and Alligator Clips -2
Optional Accessories
USB-GPIB

USB-GPIB adapter[1]The latest version of EasyDMM can be downloaded for free from the SDM3000 series of digital multimeter. Please see our web site at www.
siglent.com for more information.

www.siglent.com 107

Digital Multimeter

SDM3045X Digital Multimeter

Application• Research Laboratory• Development Laboratory• Detection and Maintenance• Calibration Laboratory• Automatic Production Test

Main Features SDM3045X• Real 4½ digit (60000 count) readings resolution• Up to 150 rdgs/s measurement speed• True-RMS AC Voltage and AC Current measuring• 1 Gb NAND flash size, Mass storage configuration files and data files• Built-in cold terminal compensation for thermocouple• With easy, convenient and flexible PC software: EasyDMM• Standard interface: USB Device, USB Host, LAN (Optioanal Accessories: USB-GPIB Adapter)• USB & LAN remote interfaces support common SCPI command set. Compatible with other popular DMMs on the market

www.siglent.com 108

Digital Multimeter
Special Features• Histogram• Trend Chart• Bar Chart• Statistics• Dual Display• Hold Measurement• dBm Hold Measurement• Interfacewww.siglent.com 109

Digital Multimeter
Ordering Information
Standard Accessories
Power Cord-1

USB Cable-1

Quick Start-1

warranty Card

-1

[1]

EasyDMMsoftware system

Test Leads and Alligator Clips -2
Optional Accessories
USB-GPIB adapter

USB-GPIB[1] The latest version of EasyDMM can be downloaded for free from the SDM3000 series of digital multimeter. Please see our web site at www.
siglent.com for more information.

www.siglent.com 110

Probes & Accessories

Probes and Accessories
Type

Model
PB470
PP510
PP215

PictureSpecifications
PB470, 70 MHz bandwidth
PP510, 100 MHz bandwidth
PP215, 200 MHz bandwidth
1 X/10 X decay, 1 M/10 Mohm, 300 V/600 V

Passive Probe
Bandwidth 250 MHz, fixed 10X decay, the rise time of about 1.2 ns,
PB925

input capacitance: 16 pF, compensation range: 10 pF-35 pF,
input impedance 10 MΩ, length 120 cm, safe voltage levels: CAT II 1000 V, CAT III
600 V

Bandwidth(-3dB) 1GHz, input capacitance 1.2 pF, input impedance 1MΩ, DC bias
SAP1000

range ±12V, probe attenuation factor ÷10, DC bias accuracy <3%, DC gain accuracy<3%, input dynamic range ±8V, non-destructive voltage range 20 V, length 130 cm

Bandwidth(-3dB) 2.5 GHz, input capacitance 1.1 pF, input impedance 1MΩ, DC bias
SAP2500

range ±12V, probe attenuation factor ÷10, DC bias accuracy <3%, DC gain accuracy<3%, input dynamic range ±8V, non-destructive voltage range 20 V, length 130 cm

Active Probe

Bandwidth(-3dB) 2.5 GHz, input capacitance 1.0 pF, input impedance 200 kohm(Diff),
100 kohm(Single ended), 50 khom(Comm mode), DC bias range ±8 V, probe
SAP2500D

attenuation factor ÷10, DC bias accuracy <3%, DC gain accuracy <3%, input dynamic
range ±8V, differential input dynamic range ±4 V, common mode input range ±10 V,
non-destructive voltage range 20 V, length 130 cm
Bandwidth(-3dB) 5 GHz, input capacitance 400 fF, input impedance 20
kohm(Diff), 10 kohm(Single ended), 50 khom(Comm mode), DC bias

SAP5000D

range ±12 V, probe attenuation factor ÷10, DC bias accuracy <3%, DC gainaccuracy<3%, input dynamic range ±8V, differential input dynamic range±2.5 V, , nondestructive voltage range 20 V, length 130 cm
Bandwidth: DC-600 kHz; Current range L, H; Maximum operation current 10 A(L)/ 100

CPL5100

A(H); Max operation voltage 600 V ; DC Accuracy: 3%±50 mA (L) ; 1500 mA~40 A
Peak: 4%±50 mA; 40 A~100 A Peak: ±15% Maximum (H); 9 V alkaline layer-built
battery/ 15 H

Bandwidth : 200 kHz; Maximum continuous current 20 Arms;
CP4020Peak current 60 A; Switching ratio: 50 mV/A; 5 mV/A;
DC measurement accuracy: 50 mV/A (0.4 A-10 ApK) ± 2%;
5 mV/A (1 A-60 ApK)±2%; 9 V battery-powered
Bandwidth: 1 MHz; Maximum continuous current 50 Arms;
Peak current 140 A; Switching ratio: 500 mV/A; 50 mV/A;

Current Probe

CP4050DC measurement accuracy: 500 mV/A (20 mA-14 ApK) ±3%±20 mA;
50 mV/A (200 mA-100 ApK )±4%± 200 mA;
50 mV/A (100 A-140 ApK)±15% max; 9V battery-powered
Bandwidth: 300 kHz; Maximum continuous current 70 Arms;

CP4070Peak current 200 A; Switching ratio: 50 mV/A; 5 mV/A;
DC measurement accuracy: 50 mV/A (0.4 A-10 ApK) ±2%,5 mV/A (1 A-200 ApK)±2%;9 V battery-powered
Bandwidth: 300 kHz; Maximum continuous current 70 Arms; Peak current 200
A;Switching ratio：100 mV/A;10 mV/A;

CP4070A

DC measurement accuracy：100 mV/A (50 mA-10 ApK) ±3%±50 mA;
10 mV/A (500 mA-40 ApK) ±4%±50 mA;
10 mV/A (40 A-200 ApK) ±15% max; 9 V battery-poweredwww.siglent.com 111

Probes & Accessories
Type

Model

PictureSpecifications
Bandwidth: 50 MHz; Maximum continuous current 30 Arms; Peak current 50

SCP5030

A;Switching ratio: 5 A/30 A;
Accuracy: 5 A(±1%±1 mA); 30 A(±1%±10 mA); Powered by oscilloscope via SAPBUS

Bandwidth: 100 MHz; Maximum continuous current 30 Arms; Peak current 50 A;
SCP5030A

Switching ratio: 5 A/30 A;
Accuracy: 5 A(±1%±1 mA); 30 A(±1%±10 mA); Powered by oscilloscope via SAPBUS

Bandwidth: 12 MHz; Maximum continuous current 150 Arms; Peak current
SCP5150300 A; Switching ratio: 30 A/150 A;
Accuracy: 30 A(±1%±10 mA); 150 A(±1%±100 mA); Powered by oscilloscope via
SAPBUS

Bandwidth: 2 MHz; Maximum continuous current 500 Arms; Peak current 750 A;
SCP5500

Switching ratio: 75 A/500 A;
Accuracy: 75 A(±1%±10 mA); 500 A(±1%±100 mA); Powered by oscilloscope via
SAPBUS

Current Probe
Bandwidth: 50 MHz; Maximum continuous current 30 Arms; Peak current

CP603050A;Switching ratio;5 A/30 A;
Accuracy: 5A(±1%±1 mA);30A(±1%±10 mA); Standard DC12V/1A power adapter

Bandwidth: 100 MHz; Maximum continuous current 30 Arms; Peak current 50 A;
CP6030A

Switching ratio: 5 A/30 A;
Accuracy: 5 A(±1%±1 mA); 30 A(±1%±10 mA); Standard DC12 V/1 A power adapter

Bandwidth: 12 MHz; Maximum continuous current 150 Arms; Peak current 300 A;
CP6150Switching ratio: 30 A/150 A;
Accuracy: 30 A(±1%±10 mA); 150 A(±1%±100 mA); Standard DC12 V/1 A power
adapter

Bandwidth: 5 MHz; Maximum continuous current 500 Arms; Peak current 750 A;
CP6500Switching ratio: 75 A/500 A;
Accuracy: 75 A(±1%±10 mA); 500 A(±1%±100 mA); Standard DC12 V/1 A power
adapter

Bandwidth: DC-50 MHz, Rise time ≤7 ns; DC Accuracy ±2%; Attenuation Ratio 50
DPB1300

X/500 X;Max Differential Test Voltage (DC + Peak AC) 50 X: ±130 V, 500 X:±1300 V; DC 12
V/1.2 A PowerHigh Voltage
Differential Probe

Bandwidth: 50 MHz; Maximum input differential voltage 800 V (DC + Peak AC); Range
DPB4080

selection (attenuation ratio):10 X/100 X; Accuracy: ±1%; Standard DC 9 V/1 A power
adapter

Bandwidth: 70 MHz; Maximum input differential voltage 1500 V (DC + Peak AC);
DPB5150

Range selection (attenuation ratio): 50 X/500 X; Accuracy: ±2%; Standard 5 V/ 1 A
USB power adapter

www.siglent.com 112

Probes & Accessories
Type

Model

PictureSpecifications
Bandwidth: 100 MHz; Maximum input differential voltage 1500 V (DC + Peak AC);

DPB5150A

Range selection (attenuation ratio): 50 X/500 X; Accuracy: ±2%; Standard 5 V/ 1 A
USB power adapter

High Voltage
Differential Probe

Bandwidth: 70 MHz; Maximum input differential voltage 7000 V (DC + Peak AC);
DPB5700

Range selection (attenuation ratio): 100 X/1000 X;
Accuracy: ±2%; Standard 5 V/1 A USB power adapter

Bandwidth: 100 MHz; Maximum input differential voltage 7000 V (DC + Peak
DPB5700AAC);Range selection (attenuation ratio): 100 X/1000 X;
Accuracy: ±2%; Standard 5 V/1 A USB power adapter

High Voltage
Probe

HPB4010

SLA1016

Bandwidth: 40 MHz; Maximum measurement voltage DC: 10 KV; AC(rms): 7 KV (sine);
AC (Vpp): 20 KV (Pulse); attenuation ratio1:1000; Accuracy: ≤3%

16 logic analyzer hardware module, suitable for SDS1000X-E 4 channel series and
SDS2000X-E series oscilloscope

Logic Probe

SPL2016

Logic Probe for SDS2000X, SDS2000X Plus and SDS5000X series, 16-channel,
500 MSa/s

Near Field Probe: H field probe sets (20 mm, 10 mm, 5 mm) , E field probe (5 mm),Near-field
Probe

SRF5030T

GPIB

USB-GPIB

300 kHz~3.0 GHz; distinguished within 10 cm range of the magnetic field; for EMI
radiation interference and the intensity detector

The USB Device interface extends into the GPIB interface, USB-GPIB adapter can moreeasily complete the task of the operation command through the GPIB, USB follow theUSB2.0 specification, GPIB follow the IEEE488.2 standard

Demo Board(STB Test Board)

Deskew Fixture

STB3

DF2001A

SigIQPro

Output signals include square waves, sine, AM, pulse, PWM, fast edge, I2C, CAN, LIN
signal etc

Supporting power analysis software for calibration phase voltage and current probes
generated during transmission

A comprehensive PC-based software for general and standards-based signals creation,
supporting Bluetooth, IoT, Custom OFDM, etc.

PC Software

SigScopeLab

A professional time-domain signal analysis and oscilloscope control software runningon the Windows operating system

www.siglent.com 113

Probes & Accessories
Type

Model
SDS1X-E-RMK

PictureSpecifications
Rackmount kit , compatible with the SDS800X HD,SDS1000X-E,SDS1000XU,SDS2000X-E model; Height 4U

Single instrument rack mount kit
SDG-RMK

19″ shelf design is compatible with the SDG800, SDG1000, SDG1000X, SDG2000X,
SDG6000X, and SDG5000 series function generators as well as the SDM3000 series of
DMMs
Rackmount kit for two intruments , compatible with the SDG800, SDG1000,

SDG-2-RMK

SDG1000X, SDG2000X, SDG5000 and SDG6000X series function generator andSDM3045X, SDM3055, SDM3065X digital multimeter

Rack Mount

SPD3000-RMK

Compatible with SPD3000X / X-E / D / S / C models.4U rack height

SDS5000X-RMKRack Mount kit for SDS5000X; Height 6U

SDS6000-RMK

Rack Mount kit for SDS6000A, SNA5000A, SSA5000A; Height 7U

SDS2000-RMK

SDS2000HD-RMK

SPS5000X-RMKRackmount kit is designed for use with only one instrument, is compatible with the
SDS2000, SDS2000X, SDS2000X Plus series Oscilloscope; Height 6U

Rack Mount kit for SDS1000X HD,SDS2000X HD,SDS3000X HD; Height 6U(exactly 260 mm)

SPS5000X EIA Standard rack, height 3U

Increase the voltage and current output capabilities to generators like the SIGLENT
SDG family.
Typical Input Impedance: 15kΩ
Input:+/- 6.5V Vpp (Gain: X1)
Amplifier

SPA1010+/- 1.3 V (Gain: X10)
Gain: Switched 10V/1V and 10V/10V
Output Voltage: 25.4 Vpp
Output Current: 1.12 A
Slew Rate: ≥ 90 V/μs
Overshoot: ≤ 4%
Compatible with all SIGLENT SDG series generators
Digital multimeter 30A current shunt-1mV/A, DC ~ 1 kHz ±0.3%, 1 kHz ~ 5 kHz ±5%,

Current Plug-in

SCD30A

30Vac RMS MAX, 60 Vdc MAX (only compatible with SDM series multimters which are
4 1/2 or above)

Attenuator

www.siglent.com 114

ATT-20 dB

20dB attenuatorProbes & Accessories
Type

Test fixture

Model

PictureSpecifications

FX-USB2

USB 2.0 test fixture

FX-ETH

100Base-TX & 1000Base-T compliance test fixture

FX-AMETH

100Base-T1 & 1000Base-T1 compliance test fixture

Output Sine, Square, Ramp, pulse, Noise, DC and 45 built-in waveforms.
USB AWG Module

SAG1021I

The arbitrary waveforms can be accessed and edited by the EasyWave PC software.
Isolated voltage ±42 Vpk.

Carry Bag

Cover

BAG-S1

Soft Carry Case for SDS1000DL+/CML+, SDS1000X, SDS1000X-E, SDS2000X-E Series

BAG-S2

Soft Carry Case for SDS2000X, SDS5000X, SSA3000X, SVA1000X, SSA3000X Plus

BAG-H1

Soft Carry Case for SHS800X/SHS1000X

FC1

Protective Cover for SNA5000A, SDS6000A

www.siglent.com 115

Other Products Overview

Other Products Overview
SIGLENT also provides other instruments like Spectrum Analyzer, Vector Network Analyzer, RF/MW Signal Generator※

SSA5000A

Spectrum Analyzer ※SSA3000X-R

SVA1000X

Frequency Range

9 kHz ~ 13.6/26.5 GHz

9 kHz ~ 3.2/5/7.5 GHz

9 kHz ~ 1.5/3.2/7.5 GHz

Real-Time Spectrum Analysis

○

√

×

Tracking Generator
Vector Network Analyzer
EMI Measurement
SSB Phase Noise
Displayed Average Noise
Level
Signal Modulation Analysis
Cable and antenna testing
Advanced Measurement Kit

×SSA3000X Plus
9 kHz ~
1.5/2.1/3.2/7.5 GHz×○SHA850A
9 kHz ~ 3.6/7.5 GHz
×

○

×

√

√

√

√

×

○

<-105 dBc/Hz<-99 dBc/Hz<-98 dBc/Hz<-98 dBc/Hz<-104 dBc/Hz-165 dBm/Hz-165 dBm/Hz-165 dBm/Hz-165 dBm/Hz-165 dBm/Hz

○

○

×

○

×

○

○

○

○

○

○

○

○

○

○

○

×

√

○

○

Remote Control Capability

√

√

√

√

√

Touch Screen

12.1’’

10.1’’

10.1’’

10.1’’

8.4’’√：Standard○：Option

www.siglent.com 116

×：Not SupportOther Products Overview※

Vector Network Analyzer
Frequency Range

Vector Network Analyzer ※SNA5000A

SHN900A

SHA850A

SVA1000X

SSA3000X-R

9 kHz ~ 4.5/8.5 GHz
100 kHz ~ 13.5/26.5 GHz

30 kHz ~14/20/26.5 GHz

100 kHz ~3.6/7.5 GHz

100 kHz ~1.5/3.2/7.5 GHz

100 kHz ~ 3.2/5/7.5 GHz

Port

2/4

2-path-2-port

1-path-2-port

1-path-2-port

1-path-2-port

Spectrum Analyzer
Frequency Range

9 kHz ~ 4.5/8.5 GHz
100 kHz ~ 13.5/26.5 GHz

30 kHz ~ 14/20/26.5
GHz

9 kHz ~ 3.6/7.5 GHz

9 kHz ~ 1.5/3.2/7.5 GHz

9 kHz ~ 3.2/5/7.5 GHz-

Level resolution

0.05 dB

0.01 dB--

Range of IFBW

10 Hz~3 MHz

10 Hz~3 MHz

10 kHz

10 kHz

10 kHz-45 dBm ~ + 10 dBm-40 dBm ~ 0 dBm-20 dBm ~ 0 dBm-20 dBm ~ 0 dBm

Setting range of output level -55 dBm ~+10 dBm
Dynamic range

Types of calibration

125 dB

100 dB114 dB

90 dB

90 dB

Response calibration,
Enhanced Response
calibration, Full-one port
calibration, Full-two port
calibration, TRL calibration

Response calibration,

Short Response,Open Response,
Full 1-Port(OSL),
Response Through,Enhanced Response

Short Response,Open Response, Full
1-Port(OSL), Response
Through, Enhanced
Response

Short Response,Open Response, Full
1-Port(OSL), Response
Through, Enhanced
Response

S11, S21

S11, S21

Enhanced Response
calibration, Full-one
port calibration, Fulltwo port calibration, TRL
calibration

Types of measurement

Scattering parameter measurement, differential
parameter measurement, receiver measurement, timedomain parameter analysis, limit test, ripple test,
S11, S21
impedance conversion, fixture simulation, adapter
removal/insertion, spectrum analysis frequency offset,
scalar mixer measurement○

TDR
Bias-Tees

√

○

√

√
√

×

×

×

×

Remote control

√

√

√

√

√

Touch screen

12.1’’

8.4’’

8.4’’

10.1’’

10.1’’√：Standard○：Option

×：Not Support

www.siglent.com 117

Other Products Overview※

RF/MW Signal Generator ※SSG6000A

SSG5000A

SSG5000X

SSG3000X

Frequency range (CW MODE)

100 kHz ~ 13.6/20/40 GHz

9 kHz ~ 13.6/20 GHz

9 kHz ~ 4/6 GHz

9 kHz ~ 2.1/3.2 GHz

Frequency range (IQ MODE)

××

10 MHz ~ 4/6 GHz

10 MHz ~ 2.1/3.2 GHzInternal modulation generator

×

×

√

×

Frequency setting resolution

0.01 Hz0.001 Hz0.001 Hz0.01 Hz

Amplitude resolution

0.01 dB

0.01 dB

0.01 dB

0.01 dB

Total amplitude accuracy≤ 0.7 dB

≤ 0.7 dB

≤ 0.7 dB

≤ 0.7 dB

SSB phase noise (offset 20 kHz@-135 dBc/Hz
1 GHz)-120 dBc/Hz-120 dBc/Hz-110 dBc/Hz

Level setting range-130 dBm ~ 24 dBm-130 dBm ~ 20 dBm-140 dBm ~ 26 dBm-110 dBm ~ 20 dBm

××

Custom digital modulation mode
ARB mode
Pulse generator
Pulse train generator
Pulse modulation
Power meter control kit

×
○
○
○
√

×
○
○
○

√
√

○
○
√

×
×
○
○
√

√

√

√

Remote control

√

√

√

√

Touch screen

5’’

5’’

5’’

5’’

√：Standard○：Option

www.siglent.com 118

×：Not Support

Service
Service Promise:
Since the date of purchase, we offer three year’s warranty for the main unit:• During the warranty period, if the products cause any hardware or software failure because of the quality, Siglent's after-sales service centeror Siglent's designated maintenance points will offer the maintenance of the fault products for the user.
• Because of improper use or any other artificial reason, the damage won’t be included in the free maintenance.

1. Extension after-sales service
Extension service is based on the main unit (not including accessories) as an object. During the extension service, Siglent still offer freemaintenance after the standard warranty period.
1.1 Three advantages:• Guarantee investment. To extend the life cycle of the products.
• Save money. To prevent the high cost of maintenance after the warranty period.
• Avoid the repeated investment. To prevent buying new equipments because it can’t be repaired after the warranty period.
1.2 The content of the extension service
You can buy the following extension service according to your demand:
Solution

Viability

ES4

One year after the warranty period

Instruction
According to the service terms, Siglent will offer
another one year for the after-sales maintenanceservice
According to the service terms, Siglent will offer

ES5

Two years after the warranty period

another two years for the after-sales maintenanceservice

2. Calibration services
After long-term use, oscilloscope will cause the deviation of measured value and waveform display, because of its work temperature andhumidity. Siglent will restore the original performance and accuracy of factory setting to calibrate the deviation.
• Eliminate the error of measurement• Restore the original performance and accuracy of the factory setting to the “new” state• The upgrade of the firmware and the software• Make the instruments comply with the standard of the ISO9001 quality management process• Traceable calibration certificateswww.siglent.com 119

About SIGLENTSIGLENT is an international high-tech company, concentrating on R&D, sales,
production and services of electronic test & measurement instruments.
SIGLENT first began developing digital oscilloscopes independently in 2002. After
more than a decade of continuous development, SIGLENT has extended its productline to include digital oscilloscopes, isolated handheld oscilloscopes, function/arbitrary
waveform generators, RF/MW signal generators, spectrum analyzers, vector network
analyzers, digital multimeters, DC power supplies, electronic loads and other general
purpose test instrumentation. Since its first oscilloscope was launched in 2005,
SIGLENT has become the fastest growing manufacturer of digital oscilloscopes. We
firmly believe that today SIGLENT is the best value in electronic test & measurement.

Headquarters:
SIGLENT Technologies Co., Ltd
Add: Bldg No.4 & No.5, Antongda Industrial
Zone, 3rd Liuxian Road, Bao'an District,
Shenzhen, 518101, China
Tel: + 86 755 3688 7876
Fax: + 86 755 3359 1582
Email: sales@siglent.com
Website: int.siglent.com
USA:
SIGLENT Technologies America, Inc
6557 Cochran Rd Solon, Ohio 44139
Tel: 440-398-5800
Toll Free: 877-515-5551
Fax: 440-399-1211
Email: info@siglent.com
Website: www.siglentna.com
Europe:SIGLENT Technologies Germany GmbH
Add: Staetzlinger Str. 70
86165 Augsburg, Germany
Tel: +49(0)-821-666 0 111 0
Fax: +49(0)-821-666 0 111 22
Email: info-eu@siglent.com
Website: www.siglenteu.com

Follow us on
Facebook: SiglentTech

modiﬁcation date：2024-05