SSA3000X
SSA3000X Plus
SSA3000X-R
SVA1000X
Spectrum Analyzer

Quick Guide
QG0703P-E02D

1

Quick Guide

Guaranty and Declaration
Copyright
SIGLENT TECHNOLOGIES CO., LTD All Rights Reserved.

Trademark Information
SIGLENT is the registered trademark of SIGLENT TECHNOLOGIES CO., LTD

Declaration

SIGLENT products are protected by patent law worldwide



SIGLENT reserves the right to modify or change parts of or all the specifications
or pricing policies at company’s sole decision.



Information in this publication replaces all previously corresponding material.



Information in this publication covers all SSA and SVA series analyzers products,
some models may have different parameters, please refer to their datasheets.



Any way of copying, extracting or translating the contents of this manual is notallowed without the permission of SIGLENT.

SIGLENT will not be responsible for losses caused by either incidental orconsequential in connection with the furnishing, use or performance of thismanual as well as any information contained.

Product Certification
SIGLENT guarantees this product conforms to the national and industrial standards in
China as well as the ISO9001: 2008 standard and the ISO14001: 2004 standard.
Other international standard conformance certification is in progress.

General Safety Summary
Carefully read the following safety precautions to avoid any personal injury or damage to the
instrument and any products connected to it. To avoid potential hazards, please use theinstrument as specified.
Use Proper AC Power Line
Only the power cord designed for the instrument and authorized by local country should be used.
Ground the Instrument
The instrument is grounded through the protective earth conductor of the power line. To avoid electric
shock, please make sure the instrument is grounded correctly before connecting its input or output
terminals.
Connect the Probe Correctly.
If a probe is used, do not connect the ground lead to high voltage since it has isobaric electric potential
as the ground.
Look Over All Terminals’ Ratings
To avoid fire or electric shock, please look over all ratings and sign instruction of the instrument. Before
connecting the instrument, please read the manual carefully to gain more information about the ratings.
Use Proper Overvoltage Protection
Make sure that no overvoltage (such as that caused by a thunderstorm) can reach the product, or elsethe operator might be exposed to danger of electrical shock.
Electrostatic Prevention
Operate the instrument in an electrostatic discharge protective area environment to avoid damages
induced by static discharge. Always ground both the internal and external conductors of the cable torelease static before connecting.
Maintain Proper Ventilation
Inadequate ventilation may cause increasing of the instrument’s temperature, which will eventually
damage the instrument. So keep well ventilated and inspect the intake and fan regularly.
Avoid Exposed Circuit or Components
Do not touch exposed contacts or components when the power is on.
Do Not Operate Without Covers
Do not operate the instrument with covers or panels removed.

Quick Guide

1

Use proper Fuse.
Use only the specified fuse.
Keep Product Surfaces Clean and Dry.
To avoid the influence of dust and/or moisture in the air, please keep the surface of the device clean
and dry.
Do Not Operate in Wet Conditions.
In order to avoid short circuiting to the interior of the device or electric shock, please do not operate theinstrument in a humid environment.
Do Not Operate in an Explosive Atmosphere.
In order to avoid damage to the device or personal injury, it is important to operate the device awayfrom an explosive atmosphere.

Safety Terms and Symbols
Terms in this Manual. These terms may appear in this manual:
WARNING
Warning statements indicate the conditions or practices that could
result in injury or loss of life.
CAUTION
Caution statements indicate the conditions or practices that could
result in damage to this product or other property.

Terms on the product. These terms may appear on the product:
DANGER

Indicates direct injuries or hazards that may happen.

WARNING Indicates potential injuries or hazards that may happen.
CAUTION

Indicates potential damages to the instrument or other property that may happen.

Symbols on the product. These symbols may appear on the product:

Hazardous

Protective Earth

Voltage

Terminal

2

Warning

Test

Power Switch

GroundQuick Guide

General Inspection
Inspect the shipping container
Keep the damaged shipping container or cushioning material until the contents of the shipment havebeen completely checked and the instrument has passed both electrical and mechanical tests.
The consigner or carrier will be responsible for damages to the instrument resulting from shipment.
SIGLENT will not provide free maintenance or replacement.
Inspect the instrument
If the instrument is found to be damaged, defective or fails in electrical or mechanical tests, pleasecontact SIGLENT.
Check the accessories
Please check the accessories according to the packing list. If the accessories are incomplete ordamaged, please contact your SIGLENT sales representative.

Preparing for Use
Appearance and Dimension

Figure 1

Quick Guide

Front View

3

Figure 2

Top View

Adjust the Supporting Legs
Adjust the supporting legs properly to use them as stands to tilt the analyzer upwards for stable
placement as well as easier operation and observation of the instrument display.

Figure 3

Before Adjusting

Figure 4

After Adjusting

Connect to AC Power Supply
The analyzer accepts 100-240V, 50/60Hz or 100-120V 400Hz AC power supply. Please use theprovided power cord to connect the instrument to the power source as shown in the figure below.
Before powering on, make sure the analyzer is protected by a fuse.

Fuse holder

The Plug

4

Figure 5

Power Cord Connection

Quick Guide

Front Panel

Figure 6

Front Panel

Table 1 Front Panel Description
NO.

Description

NO.

Description

1

User Graphical Interface, touch support

7

RF Input, VNA port 2(SVA)

2

Menu Control Keys

8

TG Output, VNA port 1(SVA)

3Function Keys

9

3.5 mm Earphone interface

4

Knob

10

USB Host

5

Numeric / Letter Keyboard

11Power Switch

6

Arrow Keys

Front Panel Function Keys

Figure 7

Quick Guide

Function Keys area5

Table 2 Function keys description (For Spectrum Analyzer Mode)Control Keys

Description

Frequency

Set the parameters of frequency, and Peak→CF, CF→Step.

Span

Set the parameters of span, and X-scale (Log-Linear) setup.

Amplitude

Set the parameters of amplitude, including Ref Level, Attenuator, Preamp, etc.; andCorrection setup.

Auto Tune

Scan the full span rapidly and move the biggest signal to center freq, andautomatically sets the optimal parameters according to the signal.

Setting Keys

DescriptionBW

Set the parameters of RBW and VBW, Average Type (Log power, Power, Voltage).

Trace

Select Trace, Trace setup and Trace math.

Sweep

Set the parameters of sweep, and EMI QPD Dwell Time.

DetectSelect the detector type for each trace independently.

Trigger

Select triggers in Free Trigger, Video Trigger and External Trigger.

Limit

Set the Pass\Fail Limit.

TG

Set the parameters of tracking generator. Including TG Level, TG Level offset
Normalization setup. The backlight LED is on when TG source is working.

Demod

Set the demodulation parameters of the AM and FM for audio listening.

Marker Keys

Description

Marker

Set the Markers and Marker Table.

Marker->Set other system parameters on the basis of the current marker value.

Marker Fn

Special functions of the marker such as noise marker, N dB bandwidth measurement
and frequency counter.

Peak

Search for the peak signal, peak search configuration and peak table.

Meas Keys

Description

Meas

In spectrum analyzer mode, selects the Advanced Measurement function.
In non-spectrum analyzer mode, select corresponding settings.

Meas Setup

Set the measurement parameters.

System Keys

Description

System

Set the system parameters.

Mode

Select the working mode between spectrum analyzer and other modes.

Display

Set the display parameters.

File

Use the file system and files.

Shortcut Keys

Description

Preset

Sets the system to certain status.

CoupleSet the parameters of some functions between auto and manual.

Help

Turn on the built-in help.

Save

Save Shortcut Key.

6

Quick Guide

Front Panel Key Backlight
The on/off state and the color of the backlights of some keys at the front panel indicate the working
state of the analyzer. The states are as listed below.

1. Power Switch
Constant on: indicates the instrument is in normal operating state.

2. ModeWhen the function is Spectrum Analyzer, the backlight turns off. When in another mode, the backlightturns on.

3. TG
When the TG source is on, the backlight of TG turns on. When the TG is output is disabled thebacklight is off.

Using the Numeric Keyboard
The analyzer provides a numeric keyboard at the front panel. The numeric keyboard supports English
uppercase/lowercase characters, numbers and common symbols (including decimal point, #, spaceand +/-) and are mainly used to edit file or folder names and set parameters.

Figure 8
1.

Numeric Keyboards+/-

In the number input mode, this key sets the sign of number. In file input mode, this toggles the character
type between numbers and letters.
2.

1 A/a

In the number input mode, this key enters the number “1”. In file input mode, this toggles the between
upper and lowercase letters.
3.

.#

In number input mode, this key enters a decimal point. In English input mode, this key enters special
characters (!, -, (, etc..).

Quick Guide

7

4.

Back

In parameter editing, press this key to delete the character to the left of the cursor.
5.

Esc



During the parameter editing process, press this key to clear the inputs in the active function area.
Press again to exit parameter input mode.

When the instrument is in remote mode (being controlled by a computer), use this key to return tolocal mode. This will unlock the front panel.

6.

Enter

In parameter editing, the system will complete the input and insert a default unit for the parameter.

Front Panel Connectors

1.

1

2

3

Figure 9

Front Panel Connectors (1)Power Switch

Power on / Power down the instrument
2.

USB Host



The analyzer can serve as a “host” device to connect an external USB devices, such as USB
storage devices, the SIGLENT USB-GPIB adapter, wireless or wired mouse and keyboard.

Save or recall functional files into/from an external USB storage device, such as saving thecurrently screenshot in .png or .jpg or .bmp format.

3.

Earphone Jack

The analyzer can demodulate AM and FM signals.Insert a 3.5 mm earphone into to the jack to acquire
the audio output of the demodulated signal. You can turn on or off the earphone output and adjust thevolume via Demod ->Volume.

CAUTION
Protect your hearing. Please turn the volume down to zero before using theearphone. Gradually turn the volume up to a comfortable level after puttingin the earphone.

8

Quick Guide

Figure 10

Front Panel Connectors

4.

TG SOURCE, VNA PORT 1

The TG SOURCE can be connected to a device-under-test (DUT) through a cable with a maleN-type male connector.

In the VNA mode, this port is used as the single port of S11 and the output port of S21.

CAUTION
To avoid damage to the tracking generator, the reverse DC voltage cannot
exceed 50 V
5.

RF INPUT, VNA PORT 2

The RF INPUT can be connected to the DUT through a cable with a male N-type connector

In the VNA mode, this port is used as the input port for S21 measurements.

CAUTION
To avoid damage to the instrument, the DC voltage component and the
maximum continuous power of the AC (RF) signal component cannot
exceed 50 V and +30 dBm respectively.

Rear Panel

Figure 11

Quick Guide

Rear Panel91.

HandlePull up the handle vertically for easy carrying of the instrument.
2.

USB Device Interface

Through this interface, the analyzer can be connected to PC for remote control.
3.

LAN Interface

Through this interface, the analyzer can be connected to the LAN for remote control.

4.

REF IN 10 MHz

The analyzer can use the internal or an external reference source.
When a 10 MHz external clock signal is received through the [10 MHz IN] connector, this signal isused as the external reference source and “Ext Ref” is displayed in the status bar of the userinterface. When the external reference is lost or not connected, the instrument switches to its
internal reference source automatically and “Ext Ref” on the screen disappears.

The [10 MHz IN] and [10 MHz OUT] connectors are usually used to build synchronization among
multiple instruments.

5.

REF OUT 10 MHz

The analyzer can use the internal or an external reference source.
When an internal reference source is used, the [10 MHz OUT] connector can output a 10 MHz
clock signal generated by the analyzer. This signal can be used to synchronize other instruments.

The [10 MHz OUT] and [10 MHz IN] connectors are usually used to build synchronization among
multiple instruments.

6.

Trigger in

In external trigger mode, the analyzer will update the trace scan after the Trigger In connector receives
an external trigger signal that meets the trigger input specifications.
7.

Security Lock HoleIf needed, you can use a security lock(purchased separately) to lock the analyzer to a desired location.
Align the lock with the lock hole and plug it into the lock hole vertically, turn the key clockwise to lock the
analyzer and then pull the key out.
8.

AC Power Supply

The analyzer accepts 100-240V, 50/60Hz or 100-120V 400Hz AC power supply. Please use the power
cord provided as accessories to connect the instrument.
9.

Fuse

Before power on, make sure the analyzer is protected by the proper input fuse.

10

Quick Guide

User Interface
1

2

3

4

5

6 7

8 9

10

11

12

13 14 15

16 17

42
41
40

43

39
38
37
36
35
34
33

44
45

32
31
30
29

28

27 26 25

24

Figure 12
Table 3

23 22

21 20

19

18

User Interface of Spectrum Analyzer Mode

Spectrum Analyzer Mode User Interface labels

NO.

Name

Description

1

SIGLENTSIGLENT logo

2

Ref

Reference level

3

UNCAL

Instrument setup is out of measurement calibration.

4

EXT REF

Valid Ext 10 MHz reference clock detected indicator

5

Att

Attenuator Value

6

Day and time

System time7

Pass/Fail status

Limit Pass/Fail status

8

MarkerCurrent active marker

9

Trace

Active trace

10

Marker instruction

current marker, touch to open a new marker

11

Marker x value

Unit: frequency, frequency delta or time

12

State indication

Auto Tune: Automatically sets the optimal parameters according to thecharacteristics of input signal
Waiting for Trigger: waiting for trigger

13

Marker y value

Amplitude value or amplitude delta value

14

USB identification

Displayed when an USB storage device is inserted

15

LAN identification

Displayed when a LAN cable is inserted

Quick Guide

11

16

Menu title

Function of the current menu.

17

Menu items

Menu items of the current function. Front panel key menus may bedifferent in different modes.

18

Operation status

Local is local mode, Remote is remote mode, Upgrademeans theinstrument is upgrading
19

Sweep progress

Indicates the currently scanned frequency position

20

Stop frequency

Stop frequency value

21

Sweep time

Time duration of a single sweep

22

Center frequency

Center frequency value

23

Span

Span value

24

VBW

Video bandwidth

25

Start frequency

The first frequency of a sweep

26

RBW

Resolution bandwidth27

Manually

When it appears, this parameter is not automatically coupled but manually

instructions

configured

Touch assistant

Click to open the commonly used functions for measurement, such as

28

peak search.
29

Trace

Trace type:

Detect type:

30

A\B\C\D

C&W: Clear Write

P-PK: Positive peak

31

status

MaxH: Max Hold

N-PK: Positive peak

MinH: Min Hold

Samp: Sample

View: View

Norm: Normal

AVG: Video average and times.

AVG: average

32

Q-PK: Quasi-peak
33

Correction

When present, indicates that there is a user-configured amplitude
correction table being mathematically applied to the displayed trace data

34

AM or FMAM or FM demodulation activated

35

PA

Enable or disable the Preamplifier

36

FFT

Sweep mode is FFT

37Single or Continue

Sweep mode single or continuous

38

Average typeLog power\Power\Voltage power

39

Trigger type

Free\Video\External trigger

40

Ref offset

34：Ref offset identification；35：Ref offset value

41

Scale/Div

Scale value

42

Scale type

Logarithm or linearity

43

Limit line

Limit Pass/Fail level

44

Trigger level

Video trigger level

45

Display line

Reference display line

12

Quick Guide

1

2

3

4

5

6

7

8 9

13
14
16

15

12

11

Figure 13

10

User Interface of Vector Network Analyzer Mode

Table 4 Vector Network Analyzer Mode User Interface labels
NO.

Name

Description

1

SIGLENTSIGLENT logo2

Calibration Status

Cor: Calibrated;

3

Port Extensions

P: Port Extensions is On

4

Marker Table

Active Marker indication,every trace 6 markers

5MarkerCurrent trace active marker

6

Marker x valueMarker frequency

7

Marker y value

Unit depend on trace format

8

Menu title

Function of the current menu.

9Menu items

Menu items of the current function

10

Stop frequency

Stop frequency value

11

PointsMeasurement points number,101~751

12

Start frequency

The first frequency of a sweep

13

Mode

Mode indication

14

Trace active

Highlight active trace indication depend on trace format

15

Trace deactive

Gray indicationd

16

Trace

Active trace

Quick Guide

Off : Correction Off;C?: Need to re-calibrate;

13

Mode
The analyzer offers a variety of operating modes that can be purchased separately. They can beselected via the Mode key:Spectrum Analyzer Mode

The default mode of the analyzer. It offers a general-purpose spectrum analyzer with a tracking
generator, and some advanced measurement.


Vector Network Analyzer ModeIn this mode, the analyzer offers a vector network analysis function and operation UI. It is able tomeasure full 1 port S11 and single direction S21 parameters.
This mode is only available on SVA1000X series.


Distance-To-Fault Mode

It is a performance verification and failure analysis tool, used for antenna and transmission line service
and maintenance. It returns VSWR versus distance, to show the effects of poor connections, damaged
cables, or faulty antennas.
This mode is only available on SVA1000X series.


Modulation Analyzer ModeIn this mode, the analyzer offers analog signal analysis (AM/FM) and digital vector signal analysis(ASK/FSK/PSK/MSK/OAM) functions. The original RF data can also be recorded to PC, from EasyVSA
software.


EMI Measurement ModeIn this mode, the analyzer provides the EMI pre-compatibility measurement function, including asequence test, a CISPR EMC filter and QP detector, and a set of EMC limit standard lines.
Real-Time Spectrum Analyzer ModeIn this mode, the analyzer offers real-time spectrum analysis with density, 3D, spectrogram, PvT, etc.

from multi-views and dimensions to monitor complex signals.
This mode is only available on SSA3000X-R series.

Front panel key menus may be different in different modes. Different modes have their own Presetfunction.
Refer to User Manual for more details about Mode.

14

Quick GuideFirmware Operation
Check System Information
Users can get the system information by press System->“System Info”, includingProduct Model, Serial and Host ID

Software Version and hardware Version



Option Information

Load Option
Refer to the procedures below to activate the options you have purchased.
1.

Press System->“System Info”->“Load Option”

2.

Enter the license key in the onscreen window. Press Enter to confirm your input and terminate thelicense key input. Or

3.

Load the .lic file provided by pressing File ->“Load” from internal memory or USB stick.

The option will be enabled after rebooting.

Firmware Upgrade
Follow this procedure to update the instrument firmware:
1.

Download the firmware package from an official SIGLENT website.

2.

Extract and copy the .ADS file into the root directory of an USB storage device.

3.

Plug the USB stick into the USB Host connector. Press System->“System Info”-> “Firmware
Update”; find the .ADS file in USB storage device.

4.

Press the ‘Load’, the analyzer will perform the update process automatically.



The upgrade process will take several minutes. When the upgrade is completed, the machine will
reboot.



Any interruption during the update process will result in update failure and system data loss. This
is not covered under the warranty and the user will bear repair costs and shipping.



Do not remove the USB storage device until the update is finished.

Quick Guide

15

Remote Control
The analyzer supports communication with computers via USB, LAN, and GPIB-USB interfaces. By
using these interfaces, in combination with programming languages and/or NI-VISA software, users
can remotely control the analyzer based on a SCPI (Standard Commands for Programmable
Instruments) compliant command set, Labview and IVI (Interchangeable Virtual Instrument), tointeroperate with other programmable instruments.
You can also remote monitor and control the analyzer in Web Browser or EasySpectrum.
For more details, refer to the “Programming Guide” or contact your nearest SIGLENT office.

Touch Operation
The analyzer has a 10.1 inch multi-touch screen and supports various gesture operations. Including:

Press or click on the upper-right-corner of the screen to enter the main menu

Swipe up and down or left and right in the waveform area to change the X-axis center coordinate
or Y-axis reference level

Perform two-points scaling in the waveform area to change the X-axis spanClick on a screen parameter or menu for parameter selection or editing;Open and drag the marker;

Use auxiliary shortcuts to perform common operations.

You can turn the touch screen function on and off via Display->’Touch Settings’.

Using Built-in Help
The built-in help system provides information about every function key at the front panel and every
menu soft key.
Press Help and the embedded help would show up.



Click on the items in the contents tree on the left to navigate to any topics interested.

Click the back or forward arrow to go to the contents just read.



Click the close sign button in the top right corner or press the Esc front-panel key to quit the help
system.

16

Quick Guide

Service and Support
SIGLENT warrants that the products that it manufactures and sells will be free from defects in materials
and workmanship for a period of three years (accessories for a period of one year) from the date ofshipment from an authorized Siglent distributor.
If the product proves defective within the respective period, SIGLENT will provide repair or replacement
as described in the complete warranty statement. To arrange for service or obtain a copy of thecomplete warranty statement, please contact your nearest Siglent sales and service office. Except asprovided in this summary or the applicable warranty statement, SIGLENT makes no warranty of any
kind, express or implied, including without limitation the implied warranties of merchantability andfitness for a particular purpose. In no event shall SIGLENT be liable for indirect, special orconsequential damages.

Troubleshooting
Before calling SIGLENT, or returning an analyzer for service, perform the quick checks listed below.
This check may eliminate the problem.
If the problem remains still, please contact SIGLENT and provide your device information in the back of
the analyzer.
1． The Power Switch

is still dark after power on:(1) Check that the power is correct / working.
(2) Check the power cord has been connected correctly(3) Check the power fuse. If a new fuse needs to be installed, please use a specified fuse.
2． The analyzer’s screen is still dark (no display) after power on:(1) Check whether the fan is running while the screen is dark, maybe the LCD cable is loose?
(2) Check whether the fan is not running while screen is dark, maybe it has failed to start up?
Do not disassemble the instrument by yourself and contact SIGLENT.
3． The control panel is unresponsive or gives a wrong response:(1) Press all the keys at the front panel to check if all of them are normal after power on.
(2) Press System ->Self Test ->Key Test to check if all the keys are working properly.
(3) If all the keys are not working, the numeric keyboard connection might be loose or the numerickeyboard is broken.
(4) If the touch screen is not working, check if the Touch is ON in Display->Touch Settings menu.
(5) Check whether the analyzer is locked in a remote control; if so, press Esc to unlock it.
Do not disassemble the instrument by yourself and contact SIGLENT.
4． The traces on the screen do not update for a long period of time:(1) Check whether the traces are in View or other status; if so, change to Clear&Write to activiate it.

Quick Guide17

(2) Verify whether all the trigger conditions have been met and whether there is a valid trigger signal
inputting.
(3) Check whether the analyzer is in a Limit test.
(4) Check whether the analyzer is in a single sweep.
(5) Check whether the current sweep time is too long.
(6) Check whether the analyzer is in a Demod listening and the Demod time is too long.
(7) Check whether the analyzer is in a EMI measurement mode, and the Sequance is not in a Scan
status.
5． Wrong measurement results or poor precision:
To calculate the system errors and check the measurement results and precision, refer to theintroductions in “Specifications”. To reach these specifications, please:(1) Check whether all the external devices are successfully connected and are working normally.
(2) Get some knowledge of the signal under measurement and set appropriate instrument parameters.
(3) Make measurements under proper conditions, for example:
Warm-up the instrument appropriately and operate the instrument under the specified environment
temperature;
Check if the Correction is ON in SA or VNA mode.
(4) Calibrate the instrument regularly to reduce or avoid errors that might occur over time.
If you need a specific calibration after the stated calibration period, contact SIGLENT or get paid
service from authorized measurement agencies.
6． System Message:
The instrument may display prompt messages, error messages or state messages according to thecurrent working status. These messages are displayed to help you to use the instrument correctly andare not instrument failures.
User system message

Message on screen

System message description (1~199)
SWT_OOR (1)Sweep time out of range

RBW_OOR(2)RBW out of range

SWT_CCOFM(3)Can't change the sweep time in FFT mode

MRKT_UNDEF(4)Undefined marker type

MRKFT_UNDEF (5)Undefined marker function type

MRKDT_UNDEF (6)Undefined marker delta pair type

MRKRT_UNDEF (7)Undefined marker read out type

TRCT_UNDEF (8)Undefined trace type

DETT_UNDEF (9)Undefined detect type

SCA_CSWL (10)Can't set the Scale/Div with linear

MRKT_IOFF (11)The marker type is OFF, please open the current marker

MRK_NDELT (12)The marker type is not Delta

MRKRT_MBST (13)The marker read out type must be set time

18

Quick Guide

MATHT_UNDEF (14)Undefined math type

XML_ANIE (15)Xml attribute node import error

XSCA_MBSLIZS (16)

X Scale must be set liner in zero span

TG_AXIS_XSCA (17)The Scale type must be logarithm when normalize

SCALE_TG_AXIS (18)

Scale type cannot be changed to linear while nomalize on

PEAK_UNFOUND (19)No peak found. Please change the search setting

IMD_FREQ_OOR (20)Frequency of intermodulation products out of range

AUTO_FAIL (21)Auto tune process failed

EXT_REF_PLUG_IN (22)

EXT ref plug in

EXT_REF_PLUG_OUT (23)

EXT ref plug out

REF_PLL_UNLOCK (24)

Ref pll unlock

SIG_NOT_STB (25)Signal is not stable enough to track

QP_RBW_OOR (26)

RBW out of range when do quasi peak scan

LAN_PLUG_IN (150)

Ethernet cable plug in

LAN_PLUG_OUT (151)

Ethernet cable plug out

IP_CONFLICT (152)

IP address conflict

IP_INVALID (153)

IP address invalid

NETM_INVALID (154)

Netmask address invalid

GWAY_INVALID (155)

Gateway address invalid

S21_NORMALIZE_DONE (183)

Normalization of S21 done

VNA_AUTO_CAL_DONE (184)Auto calibration of VNA done

Execution error (400~599)
LCF_DTFERR (400)

Load configurations failed, due to file error

Device error (600~799)
FUF_DTVERR (600)Firmware upgrade failed, due to the version error

FUF_DTRERR (601)

Firmware upgrade failed, due to the ram error

FUF_DTFERR (602)

Firmware upgrade failed, due to the file error

FUF_DTFVERR (603)

Firmware upgrade failed, due to verify the file error

FUF_DTUZFERR (604)

Firmware upgrade failed, due to unzip the file error

LIC_INVALID (605)

License is invalid!

ADC_ERROR (606)

Warning, ADC Overload!

Quick Guide

19

About SIGLENTSIGLENT is an international high-tech company, concentrating on R&D, sales,
production and services of electronic test & measurement instruments.

SIGLENT first began developing digital oscilloscopes independently in 2002.
After more than a decade of continuous development, SIGLENT has
extended its product line to include digital oscilloscopes, isolated handheld
oscilloscopes,
generators,

function/arbitrary
spectrumwaveform

analyzers,

vectorgenerators,
network

RF/MWsignalanalyzers,

digitalmultimeters, DC power supplies, electronic loads and other general purpose
test instrumentation. Since its first oscilloscope was launched in 2005,
SIGLENT has become the fastest growing manufacturer of digital
oscilloscopes. We firmly believe that today SIGLENT is the best value inelectronic test & measurement.

Headquarters:
SIGLENT Technologies Co., Ltd
Add: Bldg No.4 & No.5, Antongda Industrial
Zone, 3rd Liuxian Road, Bao'an District,
Shenzhen, 518101, China
Tel: + 86 755 3688 7876
Fax: + 86 755 3359 1582
Email: sales@siglent.com
Website: int.siglent.com

North America:
SIGLENT Technologies America, Inc
6557 Cochran Rd Solon, Ohio 44139
Tel: 440-398-5800
Toll Free: 877-515-5551
Fax: 440-399-1211
Email: info@siglentna.com
Website: www.siglentna.com

Europe:SIGLENT Technologies Germany GmbH
Add: Staetzlinger Str. 70
86165 Augsburg, Germany
Tel: +49(0)-821-666 0 111 0

Follow us on
Facebook: SiglentTech

Fax: +49(0)-821-666 0 111 22
Email: info-eu@siglent.com
Website: www.siglenteu.com

2

Quick Guide