SSA3000X Plus
Spectrum Analyzer

DataSheet

DS0703P_E04G

General Description
The SIGLENT SSA3000X Plus series spectrum analyzers are powerful and flexible tools for RF spectrum
and signal analysis. With a frequency range to 7.5 GHz, the analyzer delivers reliable automatic
measurements and multiple modes of operation: spectrum analyzer the base, optional functions include
RF power measurement, vector signal modulation analysis, reflection measurement, and EMI
measurement.
Applications include broadcast monitoring/evaluation, site surveying, S-parameter measurement,
analog/digital modulation analysis, EMI pre-compliance test, research and development, education,
production, and maintenance.

Features and Benefits
Spectrum Analyzer Frequency Range from 9 kHz up to 7.5 GHz-165 dBm/Hz Displayed Average Noise Level (Typ.)-98 dBc/Hz.@10 kHz Offset Phase Noise (1 GHz, Typ.)Level Measurement Uncertainty < 0.7 dB (Typ.)1 Hz Minimum Resolution Bandwidth (RBW)
Preamplifier Standard
Tracking Generator Standard
Analog and Digital Signal Modulation Analysis Mode (Opt.)
Reflection Measurement Kit (Opt.)
EMI Measurement Mode (Opt.)
Advanced Measurement Kit (Opt.)
10.1 inch Multi-Touch Screen, Mouse and Keyboard supported
Web Browser Remote Control on PC and Mobile Terminals and File Operation

SSA3000X Plus Analyzer DataSheet

1Models and Main index
Model

SSA3015X Plus

SSA3021X Plus

SSA3032X Plus

SSA3075X Plus

Frequency Range

9 kHz~1.5 GHz

9 kHz~2.1 GHz

9 kHz~3.2 GHz

9 kHz~7.5 GHz

Resolution Bandwidth

1 Hz~1 MHz

1 Hz~1 MHz

1 Hz~1 MHz

1 Hz~3 MHz-156 dBm/Hz-161 dBm/Hz-161 dBm/Hz-165 dBm/Hz

SSB Phase Noise< -99 dBc/Hz< -98 dBc/Hz<-98 dBc/Hz<-98 dBc/Hz

Third-order intercept+10 dBm+10 dBm+10 dBm+14 dBm

Total Amplitude Accuracy< 1.2 dB

< 0.7 dB

< 0.7 dB

< 0.7 dB

Tracking Generator

100kHz~1.5GHz

100kHz~2.1GHz

100kHz~3.2GHz

100kHz~7.5GHz

Touch Screen

Multi Touch, Mouse and Keyboard supported

Advanced Measurement

CHP, ACPR, OBW, CNR, Harmonic, TOI, Monitor

Reflection Measurement

VSWR measurement using Reflection Bridge

EMI Test

EMI Filter and Quasi-Peak Detector, Log Scale and Limit Line

Modulation Analysis

AM, FM; PM, ASK, FSK, MSK, PSK, QAM

Communication Interface

LAN, USB Device, USB Host (USB-GPIB)

Remote Control Capability

SCPI/Labview/IVI based on USB-TMC/VXI-11/Socket/Telnet

Remote Controller

NI-MAX, Web Browser, Easy Spectrum software, File Explorer

Displayed Average Noise
Level

2

SSA3000X Plus Analyzer DataSheetDesign Features
Spectrum Analyzer Mode
10.1 lnch Display with Multi-Touch Screen

Phase noise <-98 dBc/Hz@1 GHz

Minimum 1 Hz Resolution Bandwidth (RBW)ACPR in Advanced Measurement Kit-165 dBm/Hz Displayed Average Noise Level

Monitor in Advanced Measurement Kit

SSA3000X Plus Analyzer DataSheet

3

Modulation Analysis Mode
AM/FM/PM, ASK/FSK/PSK/MSK/QAM Vector Signal Modulation Analysis, EVM evaluation

EMI Measurement Mode
EMI Measurement with CISPR 16-1-1 EMI filter, Quasi-peak Detector, and pre-stored standards.

4

SSA3000X Plus Analyzer DataSheetReflection Measurement
VSWR and Return Loss measurement with Q value calculation, using External Reflection Bridge or
Directional Coupler

SSA3000X Plus Analyzer DataSheet

5

Accessories
Utility Kit

Near Field Probe Set

USB-GPIB Adaptor

6U Rack Mount

Soft Carrying Bag

Reflection Bridge

Calibration Kit

6

SSA3000X Plus Analyzer DataSheet

Specifications

Specifications are valid under the following conditions: The instrument is within the calibration period, hasbeen stored between 0 and 50°C for at least 2 hours prior to use, and has been powered on and warmed
up for at least 40 minutes. The specifications include the measurement uncertainty, unless otherwise
noted.

Specifications: All products are guaranteed to meet published specifications when operating at room
temperature (approximately 25°C), unless otherwise noted.

Typical: Performance deemed typical implies that 80 percent of the measurement results will meet the
typical published performance with a 95th percentile confidence level at room temperature (approximately
25°C). Typical performance is not warranted and does not include measurement uncertainty.

Nominal: The expected performance or design attribute.

SSA3000X Plus Analyzer DataSheet

7

Spectrum Analyzer Mode
Frequency and Time Characteristic
Frequency
SSA3015X Plus

SSA3021X Plus

SSA3032X Plus

SSA3075X Plus

Frequency range

9 kHz ~ 1.5 GHz

9 kHz ~ 2.1 GHz

9 kHz ~ 3.2 GHz

9 kHz ~ 7.5 GHz

Frequency resolution

1 Hz

Frequency Span
Range

0 Hz, 100 Hz to Max Frequency

Accuracy± Span / (number of display points - 1)

Internal Reference Source
Reference frequency

10.000000 MHz

Reference frequency± [(time since last adjustment ×frequency aging rate) + temperatureaccuracy / uncertainty

stability + initial calibration accuracy]

Initial calibration accuracy<1 ppmTemperature stability<1 ppm/year, 0 °C ~50 °C

Frequency aging rate

<0.5 ppm/first year, 3.0 ppm/20 years

MarkerMarker resolution
Marker uncertainty
Frequency Counter resolution

Span / (number of display points - 1)
± [frequency indication × reference frequency uncertainty + 10% ×
resolution bandwidth + ½ * marker resolution + 1 Hz]
0.01 Hz

0.1 HzResolution bandwidth (-3dB)

1 Hz ~ 1 MHz, in 1-3-10 sequence

1 Hz~3 MHz

Resolution filter shape factor< 4.8 : 1 (60 dB:3 dB), Gaussian-like

RBW uncertainty<5% (nom.)

Video bandwidth (-3dB)

1 Hz ~ 3 MHz, in 1-3-10 sequence

VBW uncertainty<5% (nom.)

Bandwidths

1 Hz~10 MHz

Sweep and Trigger
Sweep time
RBW

1 ms to 1500 s

1 ms to 2100 s

1 ms to 3200 s

Sweep

30 Hz ~ 1 MHz

3 kHz ~ 3 MHz

FFT

1 Hz ~ 10 kHz

1 Hz ~ 10 kHz

Sweep rule

Single, Continuous

Trigger source

Free, Video, External

External trigger

5V TTL level, Rising edge/Falling edge

8

1 ms to 7500 s

SSA3000X Plus Analyzer DataSheetAmplitude Accuracy and Range Specifications
Amplitude and Level
SSA3015X Plus

SSA3021X Plus

SSA3032X Plus

SSA3075X PlusDANL to +10 dBm, 100 kHz ~ 1 MHz, Preamp off

Measurement range

DANL to +20 dBm, 1 MHz ~ 7.5 GHz, Preamp off

Reference level-200 dBm to +30 dBm, 1 dB steps

Pre-Amplifier

20 dB (nom.)

Input attenuation

0 ~ 30 dB, 1 dB steps

Maximum input DC voltage+/- 50 VDC

Maximum average power

30 dBm, 3 minutes, fc ≥10 MHz, att > 20 dBm, preamp off

Maximum damage level

33 dBm, fc ≥ 10 MHz, att > 20 dBm, preamp off

0 ~ 50 dB, 1 dB steps

Level Display
Logarithmic level axis

1 dB to 200 dB

Linear level axis

0 to reference level

Units of level axis

dBm, dBmV, dBμV, dBμA, Volt, Watt

Number of display points

751

Number of traces

4

Trace detectors
Trace functions

Positive-peak, Negative-peak, Sample, Normal,
Average(Voltage/RMS/Video), Quasi-peak
Clear write, Max Hold, Min Hold, View, Blank, Average, Math

SSB Phase Noise
SSA3015X Plus
Offset
10 kHz
100 kHz
1 MHz

SSA3021X Plus

SSA3032X Plus

SSA3075X Plus

20 °C to 30 °C, fc = 1 GHz, Normalized to 1 Hz
-95 dBc/Hz,-95 dBc/Hz,-95 dBc/Hz,-96 dBc/Hz,-99 dBc/Hz (typ.)-98 dBc/Hz (typ.)-98 dBc/Hz (typ.)-98 dBc/Hz (typ.)-96 dBc/Hz,-96 dBc/Hz,-96 dBc/Hz,-95 dBc/Hz,-98 dBc/Hz (typ.)-97 dBc/Hz (typ.)-97 dBc/Hz (typ.)-97 dBc/Hz (typ.)-115 dBc/Hz,-115 dBc/Hz,-115 dBc/Hz,-112 dBc/Hz,-120 dBc/Hz (typ.)-117 dBc/Hz (typ.)-117 dBc/Hz (typ.)-114 dBc/Hz (typ.)

SSA3000X Plus Analyzer DataSheet

9Displayed Average Noise Level (DANL)SSA3015X Plus

SSA3021X Plus

SSA3032X Plus

SSA3075X Plus

20 °C to 30 °C, att = 0 dB, RBW = 1 Hz, sample detector, trace average > 50, TG off
100 kHz ~1 MHz
1 MHz~10 MHz
10 MHz~200 MHz

Preampoff

200 MHz~1.5 GHz-100 dBm,-107 dBm,-107 dBm,-105 dBm,-102 dBm (typ.)-111 dBm (typ.)-111 dBm (typ.)-109 dBm (typ.)-124 dBm,-132 dBm,-132 dBm,-122 dBm,-130 dBm (typ.)-136 dBm (typ.)-136 dBm (typ.)-126 dBm (typ.)-128 dBm,-137 dBm,-137 dBm,-142 dBm,-134 dBm (typ.)-141 dBm (typ.)-141 dBm (typ.)-146 dBm (typ.)-121 dBm,-135 dBm,-135 dBm,-142 dBm,-127 dBm (typ.)-139 dBm (typ.)-139 dBm (typ.)-147 dBm (typ.)-126 dBm,-126 dBm,-140 dBm,-132 dBm (typ.)-132 dBm (typ.)-145 dBm (typ.)1.5 GHz~3.2 GHz-137 dBm,

3.2 GHz~5.0 GHz-143 dBm (typ.)-136 dBm,

5.0 GHz~6.5 GHz-141 dBm (typ.)-131 dBm,

6.5 GHz~7.5 GHz
100 kHz ~1 MHz
1 MHz~10 MHz
10 MHz~200 MHz

Preampon

200 MHz~1.5 GHz
1.5 GHz~3.2 GHz
3.2 GHz~5.0 GHz
5.0 GHz~6.5 GHz
6.5 GHz~7.5 GHz

10-139 dBm (typ.)-120 dBm,-132 dBm,-132 dBm,-133 dBm,-123 dBm (typ.)-137 dBm (typ.)-137 dBm (typ.)-136 dBm (typ.)-147 dBm,-148 dBm,-148 dBm,-151 dBm,-152 dBm (typ.)-154 dBm (typ.)-154 dBm (typ.)-154 dBm (typ.)-150 dBm,-156 dBm,-156 dBm,-161 dBm,-156 dBm (typ.)-161 dBm (typ.)-161 dBm (typ.)-165 dBm (typ.)-142 dBm,-155 dBm,-155 dBm,-159 dBm,-148 dBm (typ.)-158 dBm (typ.)-158 dBm (typ.)-163 dBm (typ.)-145 dBm,-145 dBm,-159 dBm,-149 dBm (typ.)-149 dBm (typ.)-162 dBm (typ.)-157 dBm,-161 dBm (typ.)-157 dBm,-160 dBm (typ.)-155 dBm,-159 dBm (typ.)

SSA3000X Plus Analyzer DataSheetFrequency Response
SSA3015X Plus

SSA3021X Plus

SSA3032X Plus

SSA3075X Plus

20 °C to 30 °C, 30% to 70% relative humidity, att = 20 dB, relative to fc = 50 MHz
Preamp off±0.8 dB, ±0.4 dB (typ.)Preamp on±1.2 dB, ±0.6 dB (typ.)Error and Accuracy
Resolution bandwidth

Logarithmic resolution, relative to RBW = 10 kHz

switching uncertainty± 0.2 dB (nom.)

Input attenuation

20 °C to 30 °C, fc = 50 MHz, preamp off, relative to att = 20 dB

switching uncertainty± 0.5 dB
20 °C to 30 °C, fc = 50 MHz, RBW = VBW = 1 kHz, att = 20 dB, peak detector,

Absolute amplitude

95% reliability

accuracy±0.4 dB, input signal -20 dBm, Preamp off±0.6 dB, input signal -40 dBm, Preamp on

Total amplitudeaccuracy
RF input VSWR

20 °C to 30 °C, fc>100 kHz, input signal -50 dBm ~ 0 dBm, att = 20 dB,
RBW=VBW=1 kHz, peak detector, preamp off, 95% reliability±1.2 dB±0.7 dB

Att = 10 dB, fc≥1 MHz

Att = 20 dB, fc≥1 MHz<1.5 (nom.)<1.5 (nom.)

Distortion and Spurious Responses
Second harmonic

20 °C to 30 °C, fc ≥ 50 MHz, mixer level -20 dBm, att = 0 dB, preamp off

distortion (SHI)-65 dBc / +45 dBm (nom.)

Third-order intercept(TOI)
1dB gain compression
Residual response
Input related spurious

20 °C to 30 °C, fc ≥ 50 MHz, two -20 dBm tones spaced by 100 kHz, att = 0 dB,
preamp off+10 dBm (typ.)+14 dBm (typ.)20 °C to 30 °C, fc ≥ 50 MHz, att = 0 dB, preamp off> -5 dBm (nom.)> 0 dBm (nom.)

20 °C to 30 °C, input terminated = 50 Ω, att = 0 dB
< -90 dBm
20 °C to 30 °C, mixer level = -30 dBm<-65 dBc

SSA3000X Plus Analyzer DataSheet11Tracking Generator
Frequency Parameter
SSA3015X Plus

SSA3021X Plus

SSA3032X Plus

SSA3075X Plus

Frequency Range

100 kHz~1.5 GHz

100 kHz~2.1 GHz

100 kHz~3.2 GHz

100 kHz~7.5 GHz

Frequency Resolution

1 Hz, Zero Span

RBW, sweep mode

100 Hz ~ 1 MHz

3k Hz ~ 3 MHz

Output level-20 dBm~0 dBm-40 dBm~0 dBm

Output level resolution

1 dB

Output flatness+/-3 dB (nom.)

Normalization Trace

Ref A/B/C/D-> Ref trace

VSWR< 2 (nom.)

Connector and ImpendenceN-type female, 50 Ω

Average safe reverse power

Total: 30 dBm (1 W)

Maximum safe reverse level

Voltage: ±50 VDCPower Parameter

Advanced Measurement Kit
Power Measurement
CHP, Channel Power

Channel Power, Power Spectral Density

ACPR, Adjacent Channel Power

Main CH Power, Left channel power, Right channel power

Ratio
OBW, Occupied Bandwidth

Occupied Bandwidth, Transmit Frequency Error

T-Power, Time Domain Power

Zero Span Integrated Power

CNR, Carrier Noise Ratio

C/N, Noise Power

Non-Linear Measurement
Harmonic measurement

Max Harmonic number 10

TOI, Third-Order Intercept

Measure the third-order products from two tones

Spectrum Monitor Measurement
Spectrogram

Reflection Measurement
Stimulus and Measurement
SSA3015X Plus

SSA3021X Plus

SSA3032X Plus

SSA3075X Plus

Frequency Range

5 MHz~1.5 GHz

100 kHz~2.1 GHz

100 kHz~3.2 GHz

100 kHz~7.5 GHz

RBW

100 Hz ~ 1 MHz

3k Hz ~ 3 MHz

Stimulus Power-20 dBm~0 dBm-40 dBm~0 dBm

Format

VSWR, Return Loss, Reflection Coefficient

Calibration

Open Cal, Open + Short, Open + Load

Marker Function

N dB BW, Q measurement

12

SSA3000X Plus Analyzer DataSheetModulation Analyzer Mode
Common Parameter
SSA3015X Plus

SSA3021X Plus

SSA3032X Plus

SSA3075X Plus

Frequency Range

2 MHz ~ 1.5 GHz

2 MHz ~ 2.1 GHz

2 MHz ~ 3.2 GHz

2 MHz ~ 7.5 GHz

Carrier Power Accuracy±2 dB (nom.)

Carrier Power Range-30 dBm to +20 dBm (nom.)

Analog Modulation Analysis
AM
Modulation rate range
Accuracy

20 Hz to 100 kHz
1 Hz (nom.)

Modulation rate < 1 kHz< 0.1% modulation rate (nom.)

Modulation rate ≥ 1 kHz

Modulation depth range

5% to 95%

Accuracy±4% (nom.)FM
Modulation rate range
Accuracy

20 Hz to 200 kHz
1 Hz (nom.)

Modulation rate < 1 kHz< 0.1% modulation rate (nom.)

Modulation rate ≥ 1 kHz

Frequency deviation

1 kHz to 400 kHz

Accuracy±4% (nom.)PM
Modulation rate range
Accuracy

50 Hz ~ 50 kHz
1 Hz (nom.)

Modulation rate < 1 kHz< 0.1% modulation rate (nom.)

Modulation rate ≥ 1 kHz

Phase deviation

0.2 ~ 100 rad

Accuracy±4% (nom.)

SSA3000X Plus Analyzer DataSheet

13

Digital Modulation Analysis
Measurement
ASK: 2ASK;
FSK: 2,4,8,16 level;
Modulation TypeMSK: GMSK;PSK: BPSK,QPSK,OQPSK,8PSK;DPSK: DBPSK, DQPSK, D8PSK, π/4 -DQPSK, π/8 -D8PSK;
QAM: 16,32,64,128,256

Meas Length

16 to 4096

Points/Symbol

4,6,8,10,12,14,16

Symbol Rate

1 ksps to 2.5 Msps, Symbol Rate* Points/Symbol <=10 Msps

FilterMeas/Ref Filter

Nyquist, Squrt Nyquist, Gauss, Half Sine, Rectangular

Length2 to 128

Alpha/BT

Alpha 0.01 ~ 1, BT 0.01 ~ 10

Trace
IQ Meas Time, IQ Meas Spectrum,
IQ Ref Time, IQ Ref Spectrum,
Trace Data

Time, Spectrum,
Symbol Error Chart, Err Vector Time, Err Vector Spectrum,
IQ Mag Err, IQ Phase Err,

LayoutSingle, Stacked 2, Grid 1 2, Grid 2*2
Log mag, Lin mag, Real, Imag,

Trace Formats

I-Q, Constellation, I-eye, Q-eye,Wrap Phase, Unwrap Phase, Trellis eye

Symbol Error Chart
EVM (rms EVM, peak EVM), Magnitude error,
PSK/DPSK/MSK/QAM

Phase error, IQ offset, Carrier offset, SNR Quadrature error,
Gain imbalance(not support for MSK),ASKASK Error, ASK depth, carrier offset

FSK

FSK Error, Magnitude error, FSK deviation, carrier offset

14

SSA3000X Plus Analyzer DataSheet

EMI Measurement Mode
Measurement
Measurement View

Frequency scan, Meter, Signal list

Pre-compliance Sequence

Scan, Search, Meas

EMI filter RBW (-6dB)

200 Hz, 9 kHz, 120 kHz, 1MHz (following CISPR 16-1-1)

RBW uncertainty< 5% (nom.)

Detector

Peak, Voltage Average, Quasi-Peak (following CISPR 16-1-1)

Dwell time

0 us ~ 10 s

RBW/Steps

0.1, 0.3, 0.5, 1, 2, 3

Corrections

4

Limit and Trace

3

Limit Standards

EN550xx, GB9254, FCC Part15, User defined

Attenuator

0-50 dB

ReportSignal List

Frequency scale

Linear, Logarithmic

SSA3000X Plus Analyzer DataSheet

15Inputs and Outputs
Front Panel
RF inputN-type female, 50 Ω (nom.)

TG Source

N-type female, 50 Ω (nom.)

USB host

USB-A plug, version 2.0

Ear Phone Jack

3.5 mmRear PanelUSB device

USB-B plug, version 2.0

LAN10/100 Base, RJ-45

10 MHz reference output

10 MHz, >0 dBm, BNC-type female, 50 Ω (nom.)

10 MHz reference input

10 MHz, -5 to +10 dBm, BNC-type female, 50 Ω (nom.)

External trigger input

5V TTL level, BNC-type female, 10 kΩ

Remote Control
Communication Interface

LAN, USB Device, USB Host (USB-GPIB adaptor)
SCPI / Labview / IVI based on USB-TMC / VXI-11 / Socket / Telnet;
NI-MAX;

Remote Control Capability

Web Browser (HTML 5 Supported);
Easy Spectrum software;
File Explorer (FTP)16

SSA3000X Plus Analyzer DataSheetGeneral Specification
Structure
SSA3015X Plus
Weight

SSA3021X Plus

SSA3032X Plus

SSA3075X Plus

Net: 4.40 kg (9.7 lb);

Net: 4.70 kg (10.0 lb);

Shipping: 5.20 kg

Shipping: 5.50 kg

Dimensions

393 mm × 207 mm × 116.5 mm (W×H×D)

Display

TFT LCD, 1024 × 600, 10.1 inch multi-touch screen

Storage

Internal (Flash) 256 MB, external (USB storage device) 32 GByte

Working Environment
SourceAC voltage range: 100-240 V, 50/60 Hz or 100-120 V 400 Hz;

Power consumption

35 W

Temperature
Humidity
Altitude

70 W

Working temperature: 0 °C to 40 °C,
Storage temperature: -20 °C to 70 °C
0°
C to 30 °C, ≤ 95% Relative humidity
30 °C to 50 °C, ≤ 75% Relative humidity
Operating: less than 3 km

Electromagnetic Compatibility
EN 61326-1: 2013 /Class A (The active input power of the EUT is less than 75 W. According to EN

EN 61000-3-2: 2014

61000-3-2, no limits are necessary.)

EN 61000-3-3: 2013

Plt: 0.65 Pst: 1.00, dmax: 4.00 % dc: 3.00 %dt Lim: 3.30 % dt>Lim: 500ms

IEC 61000-4-2: 2008

AD ±8.0 kV, CD ±4.0 kV

IEC 61000-4-3: 2006 +80 MHz to 1000 MHz: 10V/m, 1.4 GHz to 2.0 GHz:3 V/m,

A1: 2007 + A2: 2010

2.0 GHz to 2.7 GHz:1V/m

IEC 61000-4-4: 2004 +AC Line:±2.00 kV

A1: 2010
IEC 61000-4-5: 2005

Line to Line: 1.0 kV, Line to Earth: 2.0 kV

IEC 61000-4-6: 2008

0.15-80 MHz:3 V 1 KHz 80% AM

IEC 61000-4-8: 200930 A/m, 50/60 Hz

IEC 61000-4-11: 2004

Voltage Dips:0%/0.5P; 40%/10P; 70%/25P;
Short Interruptions Test Level % UT: 0%/250P

SafetyIEC 61010-1:2010/EN 61010-1:2010
CAN/CSA-C22.2 No.61010-1:2012,
CAN/CSA-C22.2 No.61010-2-30:2012,
UL 61010-1:2012,
UL 61010-2-30:2012
RoHS
2011/65/EU

SSA3000X Plus Analyzer DataSheet

17Ordering Information
Product

Product CodeStandard Accessories

Description

Order Number

Spectrum Analyzer, 9 kHz ~ 1.5 GHz, TG standard

SSA3015X Plus

Spectrum Analyzer, 9 kHz ~ 2.1 GHz, TG standard

SSA3021X Plus

Spectrum Analyzer, 9 kHz ~ 3.2 GHz, TG standard

SSA3032X Plus

Spectrum Analyzer, 9 kHz ~ 7.5 GHz, TG standard

SSA3075X Plus

Quick Start, USB Cable, Power Cord
Advanced Measurement Kit

SSA3000XP-AMK

Utility Kit:

UKitSSA3X

N(M)-SMA(M) cable(6 GHz), N(M)-N(M) cable(6 GHz),
N(M)-BNC(F) adaptor x2, N(M)-SMA(F) adaptor x2,
10 dB 1W attenuator
N(M)-BNC(M) cable, 70cm, 2 GHz

N-BNC-2L

Common Options andN(M)-SMA(M) cable, 70cm, 6 GHz

N-SMA-6L

Accessories

N(M)-N(M) cable, 70cm, 6 GHz

N-N-6L

N(M)-SMA(M) cable, 100cm, 18 GHz

N-SMA-18L

N(M)-N(M) cable, 100cm, 18 GHz

N-N-18L

SMA(M)-SMA(M) cable, 100cm, 18 GHz

SMA-SMA-18L

USB-GPIB Adaptor

USB-GPIBSoft carrying bag

BAG-S2

6U Rack Mount Kit

SSA-RMK

Reflection Measurement

SSA3000-Refl

Reflection Bridge Kit:

RB3X25Reflection
Measurement Options

Reflection Bridge (1 MHz~2.5 GHz), N(M)-N(M) adaptors x2
50 Ω, N type Male, 4.5 GHz Economic Calibration Kit:

F503ME

Open(M), Short(M), Match(M), Through Adapter(F-F)
EMI test Options

EMI Measurement Mode

SSA3000XP-EMI

300 kHz~3 GHz Near Field Probe Kit:

SRF5030T

3 H-probes (20/10/5 mm), 1 E-probe (5 mm)
Modulation Analysis

Digital Modulation: ASK, FSK, MSK, PSK, QAM

SSA3000XP-DMA

Options

Analog Modulation: AM, FM, PM

SSA3000XP-AMA

18

SSA3000X Plus Analyzer DataSheet