SSA5000A
Spectrum Analyzer

Quick Guide
EN_01C1
Quick Guide

Guaranty and Declaration
Copyright
SIGLENT TECHNOLOGIES CO., LTD All Rights Reserved.

Trademark Information
SIGLENT is the registered trademark of SIGLENT TECHNOLOGIES CO.,LTD

Declaration⚫

SIGLENT products are protected by patent law worldwide⚫

SIGLENT reserves the right to modify or change parts of or all the
specifications or pricing policies at company’s sole decision.

⚫

Information in this publication replaces all previously corresponding
material.

⚫

Any way of copying, extracting or translating the contents of this manual
is not allowed without the permission of SIGLENT.

⚫

SIGLENT will not be responsible for losses caused by either incidental orconsequential in connection with the furnishing, use or performance of
this manual as well as any information contained.

Product Certification
SIGLENT guarantees this product conforms to the national and industrial
standards in China as well as the ISO9001: 2008 standard and the ISO14001:
2004 standard. Other international standard conformance certification is inprogress.

Contents
Important Safety Information ....................................................................... 1
General Inspection .....................................................................................11
Preparing for Use ...................................................................................... 12
Front Panel ................................................................................................ 13
Rear Panel ................................................................................................ 17
User Interface ............................................................................................ 19
Basic Control Operation ............................................................................ 21
Firmware Operation................................................................................... 23
Remote Control ......................................................................................... 25
Service and Support .................................................................................. 25
Troubleshooting ......................................................................................... 26

Important Safety Information
This manual contains information and warnings that must be followed by theuser for safe operation and to keep the product in a safe condition.

General Safety Summary
Carefully read the following safety precautions to avoid any personal injury or
damage to the instrument and any products connected to it. To avoid potential
hazards, please use the instrument as specified.

To Avoid Fire or Personal Injury.

Use Proper AC Power Line
Only the power cord designed for the instrument and authorized by local
country should be used.

Ground the Instrument
The instrument is grounded through the protective earth conductor of thepower line. To avoid electric shock, please make sure the instrument isgrounded correctly before connecting its input or output terminals.

Look Over All Terminals’ Ratings
To avoid fire or electric shock, please look over all ratings and sign instructionof the instrument. Before connecting the instrument, please read the manual
carefully to gain more information about the ratings.

SSA5000A Quick Guide

1

Equipment Maintenance and Service
When the equipment fails, please do not dismantle the machine formaintenance. The equipment contains capacitors, power supply, transformers,
and other energy storage devices, which may cause high voltage damage. Theinternal devices of the equipment are sensitive to static electricity, and directcontact is easy to cause irreparable damage to the equipment. It is necessary
to return to the factory or the company's designated maintenance organization
for maintenance.
Be sure to pull out the power supply when repairing the equipment. Live line
operation is strictly prohibited. The equipment can only be powered on whenthe maintenance is completed and the maintenance is confirmed to be
successful.

Identification of Normal State of Equipment
After the equipment is started, there will be no alarm information and error
information at the interface under normal conditions. The curve of the interface
will scan from left to right freely; if there is a button in the scanning process orthere is an alarm or error prompt, the device may be in an abnormal state. You
need to view the specific prompt information. You can try to restart the setting.
If the fault information is still in place, do not use it for testing. Contact the
manufacturer or the maintenance department designated by the manufacturer
to carry out maintenance to avoid the wrong test data caused by the use of thefault or endanger the personal safety.

Not Operate with Suspected Failures
If you suspect that there is damage to the instrument, please let qualified
service personnel check it.

2

SSA5000A Quick GuideAvoid Circuit or Wire Exposed Components Exposed
Do not touch exposed contacts or components when the power is on.

Do not operate in wet/damp conditions

Do not operate in an explosive atmosphere

Keep the surface of the instrument clean and dry

Not to use the equipment for measurements on mains circuits, not to use
the equipment for measurements on voltage exceed the voltage range
describe in the manual. The maximum additional transient voltage
cannot exceed 1300V.

The responsible body or operator should refer to the instruction manual
to preserve the protection afforded by the equipment. If the equipment isused in a manner not specified by the manufacturer, the protection
provided by the equipment may be impaired.

Any parts of the device and its accessories are not allowed to bechanged or replaced, other than authorized by the manufacturer or agent.

SSA5000A Quick Guide

3

Safety Terms and Symbols
When the following symbols or terms appear on the front or rear panel of the
instrument or in this manual, they indicate special care in terms of safety.

This symbol is used where caution is required. Refer to theaccompanying information or documents to protect against
personal injury or damage to the instrument.
This symbol warns of a potential risk of shock hazard.
This symbol is used to denote the measurement groundconnection.
This symbol is used to denote a safety ground connection.
This symbol shows that the switch is an On/Standby switch.
When it is pressed, the analyzer’s state switches between
Operation and Standby. This switch does not disconnect thedevice's power supply. To completely power off the analyzer, thepower cord must be unplugged from the AC socket after the
instrument is in the standby state.
This symbol is used to represent alternating current, or "AC".
The "CAUTION" symbol indicates a potential hazard. It calls
CAUTIONattention to a procedure, practice, or condition which may be
dangerous if not followed. Do not proceed until its conditions are
fully understood and met.
The "WARNING" symbol indicates a potential hazard. It calls
attention to a procedure, practice, or condition which, if not

WARNING followed, could cause bodily injury or death. If a WARNING is
indicated, do not proceed until the safety conditions are fully
understood and met.

4

SSA5000A Quick Guide

Working Environment
The design of the instrument has been verified to conform to EN 61010-1
safety standard per the following limits:

Environment
The instrument is used indoors and should be operated in a clean and dry
environment with an ambient temperature range.

Note: Direct sunlight, electric heaters, and other heat sources should beconsidered when evaluating the ambient temperature.

Warning: Do not operate the instrument in explosive, dusty, or
humid environments.
Ambient Temperature
Operating: 0 ℃ to +50 ℃
Non-operation: -30 ℃ to +70 ℃

Note: Direct sunlight, radiators, and other heat sources should be taken into
account when assessing the ambient temperature.

Humidity
Operating: 5% ~ 90 %RH, 30 ℃, derate to 50 %RH at 40 ℃
Non-operating: 5% ~ 95% RH

SSA5000A Quick Guide

5

Altitude
Operating: ≤ 3,048 m, 25 ℃
Non-operating: ≤ 12,191 m

Installation (overvoltage) Category
This product is powered by mains conforming to installation (overvoltage)
Category II.

Note: Installation (overvoltage) category I refers to situations where equipment
measurement terminals are connected to the source circuit. In these terminals,
precautions are done to limit the transient voltage to a correspondingly low
level.
Installation (overvoltage) category II refers to the local power distribution level
which applies to equipment connected to the AC line (AC power).

Degree of Pollution
The analyzers may be operated in environments of Pollution Degree II.

Note: Degree of Pollution II refers to a working environment that is dry andnon-conductive pollution occurs. Occasional temporary conductivity caused bycondensation is expected.

IP Rating
IP20 (as defined in IEC 60529).

6

SSA5000A Quick Guide

Cooling Requirements
This instrument relies on the forced air cooling with internal fans and ventilation
openings. Care must be taken to avoid restricting the airflow around theapertures (fan holes) at each side of the analyzer. To ensure adequate
ventilation it is required to leave a 15 cm (6 inch) minimum gap around thesides of the instrument.

CAUTION: Do not block the ventilation holes located on both sidesof the analyzer.

CAUTION: Do not allow any foreign matter to enter the analyzer
through the ventilation holes, etc.

Power and Grounding Requirements
The instrument operates with a single-phase, 100 to 240 Vrms (+/-10%) AC
power at 50/60 Hz (+/-5%), or single-phase 100 to 120 Vrms (+/-10%) AC
power at 400 Hz (+/-5%).

No manual voltage selection is required because the instrument automatically
adapts to line voltage.

Depending on the type and number of options and accessories (probes, PC
port plug-in, etc.), the instrument can consume up to 193 W of power.

Note: The instrument automatically adapts to the AC line input within the
SSA5000A Quick Guide

7

following ranges:
Voltage Range:

90 - 264 Vrms

Frequency Range: 47 - 63 Hz

90 - 132 Vrms
380 - 420 Hz

The instrument includes a grounded cord set containing a molded
three-terminal polarized plug and a standard IEC320 (Type C13) connector for
making line voltage and safety ground connection. The AC inlet groundterminal is connected directly to the frame of the instrument. For adequate
protection against electrical shock hazards, the power cord plug must be
inserted into a mating AC outlet containing a safety ground contact. Use only
the power cord specified for this instrument and certified for the country of use.

Warning: Electrical Shock Hazard!
Any interruption of the protective conductor inside or outside of theanalyzer, or disconnection of the safety ground terminal creates ahazardous situation. Intentional interruption is prohibited.

The position of the instrument should allow easy access to the socket. To makethe instrument completely power off, unplug the instrument power cord fromthe AC socket.

The power cord should be unplugged from the AC outlet if the analyzer is not to
be used for an extended period.

CAUTION: The outer shells of the front panel terminals areconnected to the instrument’s chassis and therefore to the safety
ground.

8

SSA5000A Quick GuideCleaning
Clean only the exterior of the instrument, using a damp, soft cloth. Do not use
chemicals or abrasive elements. Under no circumstances allow moisture topenetrate the instrument. To avoid electrical shock, unplug the power cord fromthe AC outlet before cleaning.

Warning: Electrical Shock Hazard!
No operator serviceable parts inside. Do not remove covers.
Refer servicing to qualified personnel

Abnormal Conditions
Do not operate the analyzer if there is any visible sign of damage or has been
subjected to severe transport stresses.

If you suspect the analyzer’s protection has been impaired, disconnect thepower cord and secure the instrument against any unintended operation.
Proper use of the instrument depends on careful reading of all instructions andlabels.

Warning: Any use of the analyzer in a manner not specified by the
manufacturer may impair the instrument’s safety protection. This
instrument should not be directly connected to human subjects orused for patient monitoring.

SSA5000A Quick Guide

9Safety Compliance
U.S. nationally recognized testing laboratory listing1.

UL 61010-1:2012/R: 2018-11. Safety Requirements for Electrical
Equipment for Measurement, Control, and Laboratory Use – Part 1:
General Requirements.

2.

UL 61010-2-030:2018. Safety Requirements for Electrical Equipment for
Measurement, Control, and Laboratory Use – Part2-030: Particularrequirements for testing and measuring circuits.

Canadian certification
1.

CAN/CSA-C22.2 No. 61010-1:2012/A1:2018-11. Safety Requirements forElectrical Equipment for Measurement, Control, and Laboratory Use –Part 1: General Requirements.

2.

CAN/CSA-C22.2 No. 61010-2-030:2018. Safety Requirements forElectrical Equipment for Measurement, Control, and Laboratory Use –Part 2-030: Particular requirements for testing and measuring circuits.

10

SSA5000A Quick Guide

General Inspection
Inspect the shipping container
Keep the damaged shipping container or cushioning material until the contentsof the shipment have been completely checked and the instrument has passed
both electrical and mechanical tests.
The consigner or carrier will be responsible for damages to the instrument
resulting from shipment. SIGLENT will not provide free maintenance orreplacement.

Inspect the instrument
If the instrument is found to be damaged, defective or fails in electrical ormechanical tests, please contact SIGLENT.

Check the accessories
Please check the accessories according to the packing list. If the accessories
areincompleteordamaged,pleasecontact

your

SIGLENT

sales

representative.

SSA5000A Quick Guide

11

Preparing for Use
Appearance and Dimension
Adjust the supporting legs properly to use them as stands to tilt the analyzer
upwards for stable placement as well as easier operation and observation of
the instrument display.

Connect to AC Power Supply
The analyzer accepts 100-240V, 50/60Hz or 100-120V 400Hz AC power
supply. Please use the provided power cord to connect the instrument to thepower source as shown in the figure below. Before powering on, make sure theanalyzer is protected by a fuse.

The Plug

Fuse holder

12

SSA5000A Quick Guide

Front Panel

3

1

4
2
5

Table 1 Front Panel Description
NO.
1

Name
LCD
ScreenDescription
12.1 inch multi-touch screen, resolution 1280*800
Stand by status: Orange

2

Power
SwitchPower on status: White
Short press: To Stand by status with current state saved
Long press: To Stand by status without current state
saved

SSA5000A Quick Guide

13

3

Function
Keys

Function and control input panel. Most of the input
functions can be performed by the touch screen
operations.
The analyzer can serve as a “host” device to connect

4

USB Host

USB memory, USB mouse and keyboard, SIGLENT
USB-GPIB adaptor, etc.
2.92 mm male connector, compatable with

5

RF Input

2.92mm/3.5mm/SMA connector mechanically.
Use torque wrench to the male part shell wihle keep thefemale part static.

CAUTION: To avoid damage to the RF input of analyzer, the DC
voltage component cannot exceed 50 V.

CAUTION: To avoid damage to the RF input of analyzer, themaximum continuous power of the AC(RF) signal cannot exceed+27 dBm with frequency larger than 10 MHz, and cannot exceed+20 dBm with frequency less than 10 MHz.

CAUTION: Short the center pin and outer shield of cables toneutralize any potential differences before connecting the cable toany sensitive input
Use proper ESD protection when working with input connectors or
sensitive boards. Basic protection includes using a 1 MOhm
resistor-isolated ESD wrist strap and ESD minimized work surfaces.
Always use properly grounded power outlets and power cords

14

SSA5000A Quick Guide

Front Panel Function Keys

Table 2 Function keys description (For Spectrum Analyzer Mode)
Name

Description

FREQ

Set frequency and span on the X axis

AMPTD

Set amplitude on the Y axis

Sweep

Set sweep and timeBW

Set RBW filter and VBW filter

Marker

Set the Markers and Peaks

Trace

Select Trace type, Detector and Trace operations.

Peak Search

Trigger

Search for the peak signal, peak search configurationand peak table.
Select triggers in Free Trigger, Video Trigger andExternal Trigger.

SSA5000A Quick Guide

15

Single/Cont

Shortcut for sweep control

Restart

Shortcut for sweep control

User Menu

Long press a menu and add it to the User Menu as aquick access

Input/Output

Set the ref clock ,correction and trigger ou, etc

Mode/Meas

Selects the modes and measurement

Meas Setup

Set the modes and measurement parameters.

System

Set the system parameters and configrations.

Preset

Sets the system to certain status.

File

Use the file browser.

Help

Turn on the built-in help.

Quick Save

Save function shortcut.

Touch

Turn on/off the touch screen.

16

SSA5000A Quick Guide

Rear Panel

1

2

3

4

5

6

7

8

9 10 11 12

13

Table 3 Rear Panel Description
NO.

Name

Description

1

HandlePull up the handle vertically for easy carrying

SafetyIf needed, you can use a security lock(purchased

Lock Holeseparately) to lock the analyzer to a desired location.

GND

Ensure the GND terminal is well connected with theTerminal

earth at work and before power on.

2

3

Through this interface, the analyzer can be
4LANconnected to your local network (LAN) for remotecontrol. Set the parameter in System menu.

5

USB

Through this interface, the analyzer can be

Deviceconnected to PC for remote control.

SSA5000A Quick Guide

17

NO.

Name

6

HDMI

Description
Output the LCD screen display to the externalmonitor through HDMI cable for synchronous display.
Audio jack, which can be externally connected with

7

Audio

headphones or speakers to monitor the audio
demodulation output .
Set the parameter in Meas Setup menu.
The OCXO reference clock source with better

8

OCXO

temperature coefficient and clock accuracy can beselected to obtain a more stable reference clock. This
module is a factory installed option.

9

REF IN

Input a 10 MHz reference clock to the analyzer.

10 MHzSet the parameter in Input/Output menu.
Output the 10 MHz reference clock of the analyzer.

10

REF OUTThe [REF IN] and [REF OUT] are usually used to

10 MHzbuild synchronization among multiple instruments.
Set the parameter in Input/Output menu.
Trigger output to indicate various working states

11

TRIG OUT

inside the analyzer, which is used to establish event
synchronization among multiple instruments.
Set the parameter in Input/Output menu.
In external trigger mode, the analyzer will update the12

TRIG IN

trace scan after the Trigger In connector receives anexternal trigger signal that meets the trigger input
specifications. Set the parameter in Trigger menu.

13

18

AC Power
SupplyBefore power on, please ensure that the ground wireof AC power supply is well connected with the earth,and the fuse works normally before power on.

SSA5000A Quick Guide

User Interface

Table 4 Spectrum Analyzer Mode User Interface
No

Name

Description

1

SIGLENTSIGLENT logo
Indicate the current working mode and2Mode/Measure

measurement function of the analyzer, and click toswitch as spectrum analysis mode, real-time
spectrum mode, etc.
Shortcuts to common used measurement

3

Shortcut Menu

functions and log bar, such as peak search,
screen capture, file browser, preset, and operation
messages.

SSA5000A Quick Guide

19

4

Instrument

Indicate the main working states of trace,

Configurationinterface, sweep, trigger, etc.
Display the signal measurement results of the5

Measurement

analyzer in various forms such as waveform,

Resultspectral line, cursor, table, statistics, constellation
diagram, etc

6

7

Sweep

Indicate and control the main sweep parameters,

Parameterssuch as frequency, resolution, scanning time, etc.

Menu

Access to all menus. Set and show all theparameters and configrations of the analyzer.
Control the opening and closing of the menu area.

8

Menu Switch

After closing, a larger measurement result area
can be displayed.

9
10

20

Interface

Indicates and controls the connection status of

StatusLAN, USB memory and other devices.

Date and Time

Indicates and controls the time and date.

SSA5000A Quick Guide

Basic Control Operation
Keyboard operation
The analyzer can be fully controlled by touch screen or mouse, while some
pop-up menus cannot be accessed by keyboard buttons. Use keyboard
buttons and their combinations for full control of menu area switching.

Touch screen and mouse operation
The analyzer provides a 12.1 inch multi-touch screen and supports various
gesture operations including:◆Press or click on the upper-right-corner of the screen to enter the mainmenu◆

Swipe up and down or left and right in the waveform area to change theX-axis center coordinate or Y-axis reference level◆

Perform two-points scaling in the waveform area to change the X-axis
span◆

Click on a screen parameter or menu for parameter selection or editing;◆Open and drag the marker;◆

Use auxiliary shortcuts to perform common operations.

You can turn the touch screen function on and off via

SSA5000A Quick Guide

Touch .

21

Using Built-in Help
The built-in help system provides information about every function key at thefront panel and every menu soft key.

◆

Press

◆

Click on the items in the contents tree on the left to navigate to any topics

Help

and the embedded help would show up.

interested.
◆

Click the back or forward arrow to go to the contents just read.

◆

Click the close sign button in the top right corner or press the Esc
front-panel key to quit the help system.

22

SSA5000A Quick GuideFirmware Operation
Check System Information
Users can get the system information by press

System> “System” >“About”, including:◆

Product Model, Serial and Host ID◆

Software Version and hardware Version◆

Option Information

Load Option
Refer to the procedures below to activate the options you have purchased.
System> “System” > “Load Option”

1.

Press

2.

Enter the license key in the onscreen window. PressEnter

to confirm

your input and terminate the license key input. Or
3.

Load the .lic file provided by pressing

File> “Load” from internal

memory or USB stick.
The option will be enabled after rebooting.

Firmware Upgrade
Follow this procedure to update the instrument firmware:
1.

Download the firmware package from an official SIGLENT website.

SSA5000A Quick Guide

23

2.

Extract and copy the .ADS file into the root directory of an USB storage
device.

3.

Plug the USB stick into the USB Host connector. Press

System>“System” > “Update” ; find the .ADS file in USB storage device.
4.

Press the ‘Load’, the analyzer will perform the update process
automatically.

CAUTION:
The upgrade process will take several minutes. When the upgrade iscompleted, the machine will reboot.
Any interruption during the update process will result in update failureand system data loss. This is not covered under the warranty andthe user will bear repair costs and shipping.
Do not remove the USB storage device until the update is finished.

24

SSA5000A Quick Guide

Remote Control
The analyzer supports communication with computers via USB, LAN, and
GPIB-USB interfaces. By using these interfaces, in combination with
programming languages and/or NI-VISA software, users can remotely control
the analyzer based on a SCPI (Standard Commands for Programmable
Instruments) compliant command set, Labview and IVI (Interchangeable
Virtual Instrument), to interoperate with other programmable instruments.
You can also remote monitor and control the analyzer in Web Browser.

For more details, refer to the “User Manual” or contact your nearest SIGLENT
office.

Service and Support
SIGLENT warrants that the products that it manufactures and sells will be freefrom defects in materials and workmanship for a period of three years(accessories for a period of one year) from the date of shipment from an
authorized Siglent distributor.
If the product proves defective within the respective period, SIGLENT will
provide repair or replacement as described in the complete warranty statement.
To arrange for service or obtain a copy of the complete warranty statement,
please contact your nearest Siglent sales and service office. Except asprovided in this summary or the applicable warranty statement, SIGLENT
makes no warranty of any kind, express or implied, including without limitation
the implied warranties of merchantability and fitness for a particular purpose. In
no event shall SIGLENT be liable for indirect, special or consequential
damages.

SSA5000A Quick Guide

25

Troubleshooting
Before calling SIGLENT, or returning an analyzer for service, perform the quick
checks listed below. This check may eliminate the problem.
If the problem remains still, please contact SIGLENT and provide your device
information in the back of the analyzer.
1.

The Power Switch

is still dark after power on:(1)

Check that the power is correct / working.

(2)

Check the power cord has been connected correctly(3)

Check the power fuse. If a new fuse needs to be installed, pleaseuse a specified fuse.

2.

The analyzer’s screen is still dark (no display) after power on:(1)

Check whether the fan is running while the screen is dark, maybe
the LCD cable is loose?

(2)

Check whether the fan is not running while screen is dark, maybe it
has failed to start up?

Do not disassemble the instrument by yourself and contact SIGLENT.

3.

The control panel is unresponsive or gives a wrong response:(1)

Press all the keys at the front panel to check if all of them are normal
after power on.

(2)

Press

System> Self Test >Key Test to check if all the keys areworking properly.
(3)

If all the keys are not working, the numeric keyboard connection
might be loose or the numeric keyboard is broken.

26

SSA5000A Quick Guide(4)If the touch screen is not working, check if the Touch is ON in
Display(5)> Touch Settings menu.

Check whether the analyzer is locked in a remote control; if so,
press

Esc

to unlock it.

Do not disassemble the instrument by yourself and contact SIGLENT.

4.

The traces on the screen do not update for a long period of time:(1)

Check whether the traces are in View or other status; if so, change
to Clear&Write to activiate it.

(2)

Verify whether all the trigger conditions have been met and whether
there is a valid trigger signal inputting.

(3)

Check whether the analyzer is in a Limit test.

(4)

Check whether the analyzer is in a single sweep.

(5)

Check whether the current sweep time is too long.

(6)

Check whether the analyzer is in a Demod listening and the Demodtime is too long.

(7)

Check whether the analyzer is in a EMI measurement mode, andthe Sequance is not in a Scan status.

5.

Wrong measurement results or poor precision:
To calculate the system errors and check the measurement results andprecision, refer to the introductions in “Specifications”. To reach these
specifications, please:(1)

Check whether all the external devices are successfully connected
and are working normally.

(2)

Get some knowledge of the signal under measurement and set

SSA5000A Quick Guide27

appropriate instrument parameters.
(3)

Make measurements under proper conditions, for example:◆

Warm-up the instrument appropriately◆

Operate the instrument under the specified environment
temperature;◆

Check if the

AMPTD> “Correction” is ON in SA or VNA

mode.
(4)

Calibrate the instrument regularly to reduce or avoid errors thatmight occur over time.
If you need a specific calibration after the stated calibration period,
contact SIGLENT or get paid service from authorized measurement
agencies.

6.

System Message:
The instrument may display prompt messages, error messages or statemessages according to the current working status. These messages are
displayed to help you to use the instrument correctly and are notinstrument failures.

Table 5 Operation Messages
User system message

Message on screen

System message description (1~199)
SWT_OOR (1)Sweep time out of range

RBW_OOR(2)RBW out of range

SWT_CCOFM(3)Can't change the sweep time in FFT
mode

MRKT_UNDEF(4)28

Undefined marker type
SSA5000A Quick GuideMRKFT_UNDEF (5)Undefined marker function type

MRKDT_UNDEF (6)Undefined marker delta pair type

MRKRT_UNDEF (7)Undefined marker read out type

TRCT_UNDEF (8)Undefined trace type

DETT_UNDEF (9)Undefined detect type

SCA_CSWL (10)Can't set the Scale/Div with linear

MRKT_IOFF (11)The marker type is OFF, please open thecurrent marker

MRK_NDELT (12)The marker type is not Delta

MRKRT_MBST (13)The marker read out type must be set
time

MATHT_UNDEF (14)Undefined math type

XML_ANIE (15)Xml attribute node import error

XSCA_MBSLIZS (16)

X Scale must be set liner in zero span

TG_AXIS_XSCA (17)The Scale type must be logarithm whennormalize

SCALE_TG_AXIS (18)

Scale type cannot be changed to linear
while nomalize on

PEAK_UNFOUND (19)No peak found. Please change the searchsetting

IMD_FREQ_OOR (20)Frequency of intermodulation products
out of range

AUTO_FAIL (21)Auto tune process failed

EXT_REF_PLUG_IN (22)

EXT ref plug in

EXT_REF_PLUG_OUT (23)

EXT ref plug out

REF_PLL_UNLOCK (24)

Ref pll unlock

SIG_NOT_STB (25)Signal is not stable enough to track

QP_RBW_OOR (26)

RBW out of range when do quasi peakscan

LAN_PLUG_IN (150)
SSA5000A Quick Guide

Ethernet cable plug in29

LAN_PLUG_OUT (151)

Ethernet cable plug out

IP_CONFLICT (152)

IP address conflict

IP_INVALID (153)

IP address invalid

NETM_INVALID (154)

Netmask address invalid

GWAY_INVALID (155)

Gateway address invalid

S21_NORMALIZE_DONE

Normalization of S21 done(183)
VNA_AUTO_CAL_DONE

Auto calibration of VNA done

(184)
Execution error (400~599)
LCF_DTFERR (400)

Load configurations failed, due to fileerror

Device error (600~799)
FUF_DTVERR (600)Firmware upgrade failed, due to theversion error

FUF_DTRERR (601)

Firmware upgrade failed, due to the ram
error

FUF_DTFERR (602)

Firmware upgrade failed, due to the fileerror

FUF_DTFVERR (603)

Firmware upgrade failed, due to verify thefile error

FUF_DTUZFERR (604)

Firmware upgrade failed, due to unzip thefile error

LIC_INVALID (605)

License is invalid!

ADC_ERROR (606)

Warning, ADC Overload!

30

SSA5000A Quick GuideAbout SIGLENTSIGLENT is an international high-tech company, concentrating on R&D, sales,
production and services of electronic test & measurement instruments.

SIGLENT first began developing digital oscilloscopes independently in 2002.
After more than a decade of continuous development, SIGLENT has extended
its

product

oscilloscopes,

line

to

includefunction/arbitrary

digitaloscilloscopes,

waveformisolated

generators,

handheld

RF/MWsignalgenerators, spectrum analyzers, vector network analyzers, digital multimeters,
DC power supplies, electronic loads and other general purpose test
instrumentation. Since its first oscilloscope was launched in 2005, SIGLENT
has become the fastest growing manufacturer of digital oscilloscopes. We
firmly believe that today SIGLENT is the best value in electronic test &measurement.

Headquarters:
SIGLENT Technologies Co., Ltd
Add: Bldg No.4 & No.5, Antongda Industrial
Zone, 3rd Liuxian Road, Bao'an District,
Shenzhen, 518101, China
Tel: + 86 755 3688 7876
Fax: + 86 755 3359 1582
Email: sales@siglent.com
Website: int.siglent.com

North America:
SIGLENT Technologies America, Inc
6557 Cochran Rd Solon, Ohio 44139
Tel: 440-398-5800
Toll Free: 877-515-5551
Fax: 440-399-1211
Email: info@siglentna.com
Website: www.siglentna.com

Europe:SIGLENT Technologies Germany GmbH
Add: Staetzlinger Str. 70
86165 Augsburg, Germany

Follow us on
Facebook: SiglentTech

Tel: +49(0)-821-666 0 111 0
Fax: +49(0)-821-666 0 111 22
Email: info-eu@siglent.com
Website: www.siglenteu.com

0

SSA5000A Quick Guide