Mechanical Calibration Kit Data Sheet

MechanicalCalibration Kit

DataSheet
EN03A0Mechanical Calibration Kit Data SheetIntroduction
Mechanical calibration kit contains individual standards to characterize systematic errors, used to calibrate
scalar or vector network analyzers.
SIGLENT mechanical calibration kit includes coaxial calibration kit and waveguide calibration kit. The
coaxial calibration kit includes termination loads, open circuits, short circuits, and through adapters, in bothsexes.

Model*1*2

Frequency

Type*3Connector

ImpendenceSimilar

F503ME

DC - 4.5 GHz

OSLT

N Type - Male

50 Ω85032B/E

F503FE

DC - 4.5 GHz

OSLT

N Type - Female

50 Ω

85032B/E

F603ME

DC - 4.5 GHz

OSLT

3.5mm - Male

50 Ω85033E

F603FE

DC - 4.5 GHz

OSLT

3.5mm - Female

50 Ω

85033E

F504MSDC - 9 GHz

OSLT

N Type - Male

50 Ω85032F

Y504MSDC - 9 GHz

OSLT

N Type - Male

50 Ω85032F

F504FSDC - 9 GHz

OSLT

N Type - Female

50 Ω

85032F

Y504FSDC - 9 GHz

OSLT

N Type - Female

50 Ω

85032F

F504TSDC - 9 GHz

OSLTN Type - Male AND Female

50 Ω

85032F

F604MSDC - 9 GHz

OSLT

3.5mm - Male50 Ω85033E

F604FSDC - 9 GHz

OSLT

3.5mm - Female

50 Ω

85033E

F604TSDC - 9 GHz

OSLT

3.5mm - Male AND Female

50 Ω

85033E

F505MSDC - 18 GHz

OSLT

N Type - Male

50 Ω85054DF505FSDC - 18 GHz

OSLT

N Type - Female

50 Ω

85054DF505TSDC - 18 GHz

OSLT

N Type - Male AND Female

50 Ω85054D

F606MS

DC – 26.5 GHz

OSLT

3.5mm - Male50 Ω85052DY606MS

DC – 26.5 GHz

OSLT

3.5mm - Male50 Ω85052DF606FS

DC – 26.5 GHz

OSLT

3.5mm - Female

50 Ω85052DY606FS

DC – 26.5 GHz

OSLT

3.5mm - Female

50 Ω85052DF606TS

DC – 26.5 GHz

OSLT

3.5mm - Male AND Female

50 Ω85052D

KWR42A

18 – 26.5 GHz

Waveguide

2.92mm-Male AND Female

50 ΩK11644A*1: Mechanical coaxial calibration kit naming rule
F/Y/S
5/6/7/8/9
0/1
3/4/5/6/7/8/9
M/F/T
E/S

Separate/Integrated/ElectricalN/3.5/2.92/2.4/1.85 mm
50/75 Ohm
4.5/9/18/26.5/40/50/67 GHz
Male/Female/BothEconomy/Standard

Mechanical Calibration Kit Data Sheet1*2: Mechanical Waveguide calibration kit naming ruleBandWVUQRKPXEIAWR10WR15
WR19WR22WR28
WR42WR62
WR90

VersionAAAAAAAAFrequency range
75 to 110 GHz
50 to 75 GHz
40 to 60 GHz
33 to 50 GHz
26.5 to 40 GHz
18 to 26.5 GHz
12.4 to 18 GHz
8.2 to 12.4 GHz*3: OSLT = Open + Short + 50Ω termination Load + Through Adapter

2Mechanical Calibration Kit Data SheetF503 Series
The F503ME and F503FE economy 50Ω N type coaxial mechanical calibration kit include terminationloads, open circuits, short circuits, and through adapters, specified from DC to 4.5 GHz.
The F503 series performance specifications are very similar to the Keysight 85032B/E mechanicalcalibration kit and it can be used as an approximate replacement of 85032B/E, or use the STD of 85032B/E
in network analyzers.

PerformanceModelF503ME

F503FE

TypeConnector

Specification

OpenN - MaleDC – 4.5 GHz, Phase Deviation* ≤ ± 0.8°

ShortN - MaleDC – 4.5 GHz, Phase Deviation ≤ ± 0.8°

LoadN - MaleDC – 4.5 GHz, SWR ≤ 1.02

AdapterN - Male to
N - Male

DC – 3 GHz, SWR ≤ 1.03
3 – 6 GHz, SWR ≤ 1.05
6 – 9 GHz, SWR ≤ 1.08

Open

N - Female

DC – 4.5 GHz, Phase Deviation ≤ ± 0.8°

Short

N - Female

DC – 4.5 GHz, Phase Deviation ≤ ± 0.8°

Load

N - Female

DC – 4.5 GHz, SWR ≤ 1.02

AdapterN - Female toN - Female

DC – 3 GHz, SWR ≤ 1.03
3 – 6 GHz, SWR ≤ 1.05
6 – 9 GHz, SWR ≤ 1.08

* Relative error to the standard phase
General Specification
ImpendencePowerInterfaces StandardDurability
TorqueSpanner
Temperature

50 Ω≤1W
IEC 60169-16 Grade 0> 2000
1.35 Nm
19 mm+ 15 °C ~ + 35 °C

Mechanical Calibration Kit Data Sheet

3F603 Series
The F603ME and F603FE economy 50Ω 3.5mm type coaxial mechanical calibration kit include
terminations loads, open circuits, short circuits, and through adapters, specified from DC to 4.5 GHz.
The F603 series performance specifications are very similar to the Keysight 85033E mechanical calibration
kit and it can be used as an approximate replacement of 85033E, or use the STD of 85033E in network
analyzers.

Performance
Model

F603ME

F603FE

TypeConnector

Specification

Open

3.5mm - MaleDC – 4.5 GHz, Phase Deviation* ≤ ± 0.8°

Short

3.5mm - MaleDC – 4.5 GHz, Phase Deviation ≤ ± 0.8°

Load

3.5mm - MaleDC – 4.5 GHz, SWR ≤ 1.028

Adapter

3.5mm - Male to3.5mm - MaleDC – 6 GHz, SWR ≤ 1.04
6 – 9 GHz, SWR ≤ 1.06
9 – 26.5 GHz, SWR ≤ 1.1

Open

3.5mm - Female

DC – 4.5 GHz, Phase Deviation ≤ ± 0.8°

Short

3.5mm - Female

DC – 4.5 GHz, Phase Deviation ≤ ± 0.8°

Load

3.5mm - Female

DC – 4.5 GHz, SWR ≤ 1.028

Adapter

3.5mm - Female to3.5mm - Female

DC – 6 GHz, SWR ≤ 1.04
6 – 9 GHz, SWR ≤ 1.06
9 – 26.5 GHz, SWR ≤ 1.1* Relative error to the standard phase
General Specification
ImpendencePowerInterfaces Standard
Durability
TorqueSpanner
Temperature

450 Ω
≤1W
IEEE Std 287> 20000.9 Nm
8 mm+ 15 °C ~ + 35 °C

Mechanical Calibration Kit Data SheetF504 and Y504 Series
The F504MS and F504FS 50Ω N type coaxial mechanical calibration kit include termination loads, opencircuits, short circuits, and through adapters, specified from DC to 9 GHz. The F504TS is a coaxial
calibration kit consisting of F504MS and F504FS.
Y504MS shares the same parts and specs as F504MS, but in integrated exterior. So does Y504FS andF504FS.
The F504 and Y504 series performance specifications are very similar to the Keysight 85032F mechanicalcalibration kit and it can be used as an approximate replacement of 85032F, or use the STD of 85032F innetwork analyzers.

Performance
ModelF504MS/Y504MS

F504TS

F504FS/
Y504FS

TypeOpen
Short
LoadConnector
N - MaleN - MaleN - MaleAdapter

N - Male to
N - Male

OpenShort
LoadAdapter

N - Female
N - Female
N - Female
N - Female
toN - Female

AdapterN - male toN - Female

WrenchN - 19mm

Specification
DC – 9 GHz, Phase Deviation* ≤ ± 0.8°
DC – 9 GHz, Phase Deviation ≤ ± 0.8°
DC – 9 GHz, SWR ≤ 1.025
DC – 9 GHz, SWR ≤ 1.06 (Return Loss ≥ -31 dB),
9 – 18 GHz, SWR ≤ 1.1 (Return Loss ≥ -26 dB),
Insert Loss ≤ 0.2 dB, Delay= 197.1 ps
DC – 9 GHz, Phase Deviation ≤ ± 0.8°
DC – 9 GHz, Phase Deviation ≤ ± 0.8°
DC – 9 GHz, SWR ≤ 1.025
DC – 9 GHz, SWR ≤ 1.06 (Return Loss ≥ -31 dB),
9 – 18 GHz, SWR ≤ 1.1 (Return Loss ≥ -26 dB),
Insert Loss ≤ 0.15 dB, Delay= 136.2 ps
DC – 9 GHz, SWR ≤ 1.06 (Return Loss ≥ -31 dB),
9 – 18 GHz, SWR ≤ 1.1 (Return Loss ≥ -26 dB),
Insert Loss ≤ 0.15 dB
1.35 Nm* Relative error to the standard phase
General Specification
ImpendenceInterfaces StandardTorqueTemperature

50 Ω
IEC 60169-16 Grade 01.35 Nm+ 15 °C ~ + 35 °C

Mechanical Calibration Kit Data Sheet

PowerDurabilitySpanner≤1W> 2000
19 mm

5F505 Series
The F505MS and F505FS 50Ω N type coaxial mechanical calibration kit include termination loads, opencircuits, short circuits, and through adapters, specified from DC to 18 GHz. The F505TS is a coaxial
calibration kit consisting of F505MS and F505FS.
The F505 series performance specifications are very similar to the Keysight 85054D mechanical calibration
kit and it can be used as an approximate replacement of 85054D, or use the STD of 85054D in network
analyzers.

Performance
Model

F505MS

TypeOpen
Short
Load
Adapter

F505TS
F505FS

Open
Short
Load
Adapter

Adapter
WrenchConnector
N - MaleN - MaleN - MaleN - Male to
N - MaleN - Female
N - Female
N - Female
N - Female toN - Female
N - male toN - FemaleN - 19mm

Specification
DC – 18 GHz, Phase Deviation* ≤ ± 1°
DC –18 GHz, Phase Deviation ≤ ± 1°
DC – 18 GHz, SWR ≤ 1.048 (Return Loss ≥ -32.6 dB)
DC – 18 GHz, SWR ≤ 1.06 (Return Loss ≥ -30.7 dB),
Insert Loss ≤ 0.2 dB, Delay= 197.1 ps
DC – 18 GHz, Phase Deviation ≤ ± 1°
DC – 18 GHz, Phase Deviation ≤ ± 1°
DC – 18 GHz, SWR ≤ 1.048 (Return Loss ≥ -30.7 dB)
DC – 18 GHz, SWR ≤ 1.06 (Return Loss ≥ -30.7 dB),
Insert Loss ≤ 0.15 dB, Delay= 136.2 ps
DC – 18 GHz, SWR ≤ 1.06 (Return Loss ≥ -30.7 dB),
Insert Loss ≤ 0.15 dB
1.35 Nm* Relative error to the standard phase

General Specification
ImpendenceInterfaces Standard
TorqueTemperature650 ΩIEC 60169-16
1.35 Nm+ 15 °C ~ + 35 °C

PowerDurabilitySpanner≤1W> 2000
19 mm

Mechanical Calibration Kit Data SheetF604 Series
The F604MS and F604FS 50Ω 3.5mm type coaxial mechanical calibration kit include termination loads,
open circuits, short circuits, and through adapters, specified from DC to 9 GHz. The F604TS is a coaxial
calibration kit consisting of F604MS and F604FS.
The F604 series performance specifications are very similar to the Keysight 85033E mechanical calibration
kit and it can be used as an approximate replacement of 85033E, or use the STD of 85033E in network
analyzers.

Performance
ModelF604MS

F604TS
F604FS

TypeOpen
Short
LoadConnector
3.5mm - Male
3.5mm - Male
3.5mm - Male

Adapter

3.5mm - Male to3.5mm - MaleOpen
Short
Load
Adapter

3.5mm - Female
3.5mm - Female
3.5mm - Female
3.5mm - Female
to3.5mm - Female

Adapter

3.5mm - male to3.5mm - Female

Wrench3.5mm - 8mmSpecification
DC – 9 GHz, Phase Deviation* ≤ ± 0.8°
DC – 9 GHz, Phase Deviation ≤ ± 0.8°
DC – 9 GHz, SWR ≤ 1.02
DC – 6 GHz, SWR ≤ 1.04
6 – 9 GHz, SWR ≤ 1.06
9 – 26.5 GHz, SWR ≤ 1.1
DC – 9 GHz, Phase Deviation ≤ ± 0.8°
DC – 9 GHz, Phase Deviation ≤ ± 0.8°
DC – 9 GHz, SWR ≤ 1.02
DC – 9 GHz, SWR ≤ 1.06 (Return Loss ≥ -31dB),
9 – 26.5 GHz, SWR ≤ 1.1 (Return Loss ≥ -26dB),
Insert Loss ≤ 0.1 dB, Delay= 83.0 ps
DC – 6 GHz, SWR ≤ 1.04
6 – 9 GHz, SWR ≤ 1.06
9 – 26.5 GHz, SWR ≤ 1.10.9 Nm* Relative error to the standard phase
General Specification
ImpendenceInterfaces StandardTorqueTemperature

50 ΩIEEE Std 287
0.9 Nm+ 15 °C ~ + 35 °C

Mechanical Calibration Kit Data Sheet

PowerDurabilitySpanner≤1W> 20008 mm

7F606 and Y606 Series
The F606MS and F606FS 50Ω 3.5mm type coaxial mechanical calibration kit include termination loads,
open circuits, short circuits, and through adapters, specified from DC to 26.5 GHz. The F606TS is a coaxial
calibration kit consisting of F606MS and F606FS.
Y606MS shares the same parts and specs as F606MS, but in integrated exterior. So does Y606FS andF606FS.
The F606 and Y606 series performance specifications are very similar to the Keysight 85052D mechanicalcalibration kit and it can be used as an approximate replacement of 85052D, or use the STD of 85052D innetwork analyzers.

Performance
Model

F606MS/Y606MSTypeOpen
Short
Load
Adapter

F606TS

F606FS/Y606FS

Open
Short
Load
Adapter

Adapter
WrenchConnector
3.5mm - Male
3.5mm - Male
3.5mm - Male
3.5mm - Male to3.5mm - Male
3.5mm - Female
3.5mm - Female
3.5mm - Female
3.5mm - Female
to3.5mm - Female
3.5mm - male to3.5mm - Female
3.5mm - 8mmSpecification
DC – 26.5 GHz, Phase Deviation* ≤ ± 1.5°
DC – 26.5 GHz, Phase Deviation ≤ ± 1.5°
DC – 26.5 GHz, SWR ≤ 1.04
DC – 26.5 GHz, SWR ≤1.06
DC – 26.5 GHz, Phase Deviation ≤ ± 1.5°
DC – 26.5 GHz, Phase Deviation ≤ ± 1.5°
DC – 26.5 GHz, SWR ≤ 1.04
DC – 26.5 GHz, SWR ≤ 1.06

DC – 26.5 GHz, SWR ≤ 1.06
0.9 Nm* Relative error to the standard phase
General Specification
ImpendenceInterfaces Standard
TorqueTemperature

850 ΩIEEE Std 2870.9 Nm+15 °C ~ + 35 °C

PowerDurabilitySpanner≤ 0.5 W
> 20008 mm

Mechanical Calibration Kit Data SheetKWR42A
The KWR42A precise K-band waveguide mechanical calibration kit contains K-band load, K-band short,
1/8λ waveguide line, 1/4λ waveguide line and 3/8λ waveguide line, specified from 17.6 GHz to 26.7 GHz.
For measurement convenience, the KWR42A includes 2.92mm coax-to-waveguide converters and somefasteners like screws, nuts, nut collars, position bolts, etc.
The KWR42A performance specifications are very similar to the Keysight K11644A mechanical calibration
kit and it can be used as an approximate replacement of K11644A, or use the STD of K11644A in network
analyzers.

Performance
Model

KWR42A

Waveguide
Waveguide

F min
(MHz)
14047
14047

F max
(MHz)
28094
28094

1/8λ Line

Waveguide

14047

28094

1/4λ Line

Waveguide

14047

28094

3/8λ Line

Waveguide

14047

28094

TypeConnector

Short
Load

Coax-towaveguideconverter

Fastener

Mechanical Calibration Kit Data Sheet

Specification
Delay = 0, Loss = 0
Delay = 0, Loss = 0
Delay = 0.751E-11 Sec,
Loss = 2.75 Gohm/Sec
Delay = 1.502E-11 Sec,
Loss = 2.75 Gohm/Sec
Delay = 2.253E-11 Sec,
Loss = 2.75 Gohm/Sec

2.92mm Female
14047 28094 VSWR≤1.25;IL≤0.5dB
to Waveguide
2.92mm - Male
14047 28094 VSWR≤1.25;IL≤0.5dB
to Waveguide
Screw M3*12, Screw M3*16, Screw M3*20, Nut M3, Nut collar M3,
Position bolt

9

General Specification
ImpendenceInterfaces Standard
TorqueTemperature

1050 Ω
IEC 60169-23
0.9 Nm+15 °C ~ + 35 °C

PowerDurabilitySpanner≤ 0.5 W
> 20008 mmMechanical Calibration Kit Data SheetCalibration Kit Definitions
ModelC0C1C2C3L0L1L2L3DelayLoss

F(e-15)F(e-27)/HzF(e-36)/Hz^2

F(e-45)/Hz^3

H(e-12)H(e-24)/HzH(e-33)/Hz^2

H(e-42)/Hz^3(pSec)(Gohm/Sec)

62.14-143.07

82.92

0.76

17.4

0.7

17.8

2.1002

Load

00.7

Thru

00.7

00.7

0.093

0.7

Load

00.7

Thru

00.7

29.2

2.2

31.8

2.36

Load02.3

Thru

02.3

29.2

2.3

31.8

2.36

Load00Thru

02.3

40.856

0.9345.955

1.087

TypeOpen

F503ME

Short

Open
F503FE-36.955

26.258049.433-310.1323.168-310.1323.1682.076589.9392536.8-264.990000-108.54

2.1705-0.01-.0.15966

Short

Open0-0.15966
2.076549.43305.5136

Short

Open
F603FE

119.09Short

Open
F603ME

0-108.54

2.1705

-0.01

13.4

F504MS

Short

Y504MS

Load

00Thru00

Mechanical Calibration Kit Data Sheet

3.3998-496.4808

34.8314-0.784711

ModelType
Open

C0C1C2C3L0L1L2L3Delay

Loss

F(e-15)F(e-27)/HzF(e-36)/Hz^2F(e-45)/Hz^3H(e-12)H(e-24)/HzH(e-33)/Hz^2H(e-42)/Hz^3(pSec)(Gohm/Sec)

89.9392536.8-264.99

13.4

41.17

0.93

45.955

1.087

F504FS

ShortY504FS

Load

00Thru00

57.9930.93

63.0781.1273Load

00Thru02.2

22.905

0.93

27.99

1.3651

Load

00Thru

02.2

29.2

2.2

31.8

2.36

Load

02.3

Thru

02.3

29.2

2.2

31.8

2.36

Load

00Thru

02.3

29.2

2.2

31.8

2.36

Open
F505MS

0.7653104.13-1943.4008

144.62-310.13

23.168-310.1323.168Open

Y606MS

Thru-310.1323.168459.8799-52.4291.5846606.2089-68.405

2.0206-108.54

2.1705-0.01-0.15966
2.076549.433-0.7847-0.15966
2.076549.43334.83142.2258-0.131549.433-496.4808

13.4

Short

F606MS

12-264.99

Short

Open
F604FS

2536.7999

Short

Open
F604MS89.939Short

Open
F505FS

3.3998-108.54

2.1705-0.01-0.15966
2.0765-108.54

2.1705-0.01

Mechanical Calibration Kit Data SheetModelC0C1C2C3L0L1L2L3DelayLoss

F(e-15)F(e-27)/HzF(e-36)/Hz^2F(e-45)/Hz^3H(e-12)H(e-24)/HzH(e-33)/Hz^2

H(e-42)/Hz^3(pSec)(Gohm/Sec)

Load

00Thru

00

29.2

2.2

31.8

2.36

TypeOpen

49.433-310.1323.168-0.15966

F606FS

Short

Y606FS

Load

00Thru000000

0.751

2.75

1.502

2.75

2.253

2.75

Short
Load
1/8λ
KWR42A

Line
1/4λ
Line
3/8λ
Line

Mechanical Calibration Kit Data Sheet

2.07650-108.54

02.1705

0-0.01013

About SIGLENTSIGLENT is an international high-tech company, concentrating on R&D, sales,production and services of electronic test & measurement instruments.
SIGLENT first began developing digital oscilloscopes independently in 2002.
After more than a decade of continuous development, SIGLENT has extended
its product line to include digital oscilloscopes, isolated handheld
oscilloscopes,

function/arbitrary

waveform

generators,

RF/MWsignalgenerators, spectrum analyzers, vector network analyzers, digital multimeters,
DC power supplies, electronic loads and other general purpose test
instrumentation. Since its first oscilloscope was launched in 2005, SIGLENT
has become the fastest growing manufacturer of digital oscilloscopes. We
firmly believe that today SIGLENT is the best value in electronic test &measurement.

Headquarters:
SIGLENT Technologies Co., Ltd
Add: Bldg No.4 & No.5, Antongda IndustrialZone, 3rd Liuxian Road, Bao'an District,Shenzhen, 518101, China
Tel: + 86 755 3688 7876
Fax: + 86 755 3359 1582
Email: sales@siglent.com
Website: int.siglent.com

North America:
SIGLENT Technologies America, Inc
6557 Cochran Rd Solon, Ohio 44139

Tel: 440-398-5800
Toll Free: 877-515-5551
Fax: 440-399-1211
Email: info@siglentna.com
Website: www.siglentna.com

Europe:SIGLENT Technologies Germany GmbH
Add: Staetzlinger Str. 7086165 Augsburg, Germany
Tel: +49(0)-821-666 0 111 0
Fax: +49(0)-821-666 0 111 22

Email: info-eu@siglent.com
Website: www.siglenteu.com

Follow us on
Facebook: SiglentTech