APU sim
The Clever Sample Preparation
System According to Column
MethodAPU sim – Your Benefits:■■ Simultaneous enrichmentof up to 6 (3 x 2) samples
according to DIN EN ISO9562■■ Extremely robust andvery good particle handlingby avoiding conventional
valve technology and using
high-quality materials■■ Intuitive operation and 		flexibility in processing■■ Minimum maintenance effort

Sample preparation for AOX analysis
limits the total throughput of AOX
samples in your laboratory? APU simoffers the perfect solution for yourdemands: thanks to simultaneousenrichment up to six samples areprepared and available for your AOX
analyzer in next to no time.
Designed with three enrichment
channels the system allows
comfortable simultaneous preparationof several samples. Each channel
takes two samples (columns) and canbe operated and started individually.
Sample volume, rinse volume and eventhe adsorption rate can be selected
separately for each channel, if required.

The flexibility of the APU sim leaves
nothing to be desired - just tailor thesample preparation to your needs.
An intuitive control panel guides you
through the clearly arranged menu
levels and makes operation child’s play.
Simply pick up or fill the sample,
press start, and let the APU sim do
the work. Whether clear or turbid
water, the system copes with every
matrix and stands out with a very
good particle handling. Thanks to
the large diameters of the hoses andthe absence of conventional valve
technology, particles are transferred
easily to the enrichment columns. The
fully automatic rinse provides freeselection of rinsing volumes.

By the Way
Columns of any size and brandcan be easily adapted to thesystem. That means APU sim
can be used as sample preparation system for a wide range ofAOX analyzers.

APU sim – Technical Parameters■■ System for sample preparation of AOX samples according to column method■■ Simultaneous processing of 6 samples in 3 separate enrichment channels■■ Sample volume, rinse volume and adsorption rate freely selectable for each channel■■ Sample volume: 5 ml to 100 ml■■ Rinse volume (nitrate wash solution): 0 ml to 100 ml■■ Adsorption rate: 1 ml / min to 6 ml / min, standard setting according to DIN EN ISO 9562: 3 ml / min■■ Semi-automatic processing: Sample filling is done manually, adsorption on activated carbon and rinsing process are fully

automatic■■ Intuitive one-button operation (Turn & Press) and clear display with symbols and numbers■■ Very good particle handling due to large system diameters and no conventional valve technology■■ Suitable for duplex and triplex columns (for samples with high particle loads) as well as for columns of all types up to a

total length of 150 mm■■ Low maintenance, recommended change of pumping hoses 1 x per year■■ Sample preparation system applicable for any commercially available AOX analyzer

High-quality materials and workmanship make the APU sim a long-lasting partner in your laboratory.
Speed up, increase your AOX sample throughput with a perfect sample preparation system, APU sim, and save TIME andMONEY!

Analytik Jena AG
Konrad-Zuse-Str. 1
07745 Jena . Germany
Phone +49 3641 77 70
Fax+49 3641 77 9279
info@analytik-jena.com
www.analytik-jena.com

Pictures: Analytik Jena AG, p. 1: iStockphoto®/Onur Döngel
Subjects to changes in design and scope of delivery as well as further technical development!

en · 10/2016 © Analytik Jena AGHeadquarters