SIGLENT Probe

Data Sheett
EN01E

SIGLENT Probe Data Sheet

Passive Probe
Model

PB470

Parameter

PP510

PP215

SP2035

Attenuation Rate

1X/10X

1X/10X

1X/10X

1X/10X

Bandwidth

10X: DC-70MHz

10X: DC-100MHz

10X: DC-200MHz

350MHzInput Impedance

1MΩ/10MΩ

1MΩ/10MΩ

1MΩ/10MΩ

1MΩ/10MΩ±2%

Input Capacitance

10X: 13pF-17pF

10X: 13pF-17pF

10X: 13pF-17pF

1X: 85pF-120pF
10X: 17pF-20pF

Compensation Range

10pF-35pF

10pF-35pF

10pF-35pF

10pF-35pF

Input Voltage

1X: 150V RMS CAT II 1X: 150V RMS CAT II 1X: 150V RMS CAT II 1X: 150V RMS CAT II
10X: 300V RMS CAT II 10X: 300V RMS CAT II 10X: 300V RMS CAT II 10X: 300V RMS CAT IIOperation Temp

-10℃ ~ 55℃-10℃ ~ 55℃-10℃ ~ 55℃-10℃ ~ 55℃

Cable Length

130 cm130 cm130 cm130 cmWeight55g

55g

55g

About 55g

Model

SP2035A

Parameter

SP3050A

SP3150A

SP5035A

Attenuation Rate

10X

10X

10X

10X

Bandwidth

350MHz500MHz500MHz350MHzInput Impedance

10MΩ±2%

10MΩ

10MΩ

10MΩ

Input Capacitance

12pF

11pF

11pF

12pF

Compensation Range

9pF-25pF

8pF-20pF

8pF-20pF

12pF-22pF

Input Voltage

10X: 300V RMS CAT II 400V RMS CAT II

400V RMS CAT II

300V RMS CAT II

Operation Temp

-10℃ ~ 55℃

0℃ ~ 50℃

0℃ ~ 50℃-10℃ ~ 55℃

Cable Length

130 cm ± 2 cm

120 cm120 cm130cm

WeightAbout 55g

55g

55g

55g

int.siglent.com

1

SIGLENT Probe Data Sheet

ModelParameter

SP5050A

PB925

Attenuation Rate

10X

10X

Bandwidth

500MHz250MHzInput Impedance

10MΩ

10MΩ

Input Capacitance

12pF

16pF

Compensation Range

12pF-22pF

10pF-35pF

Input Voltage

300V RMS CAT II600 V CAT III
1000 V CAT II

Operation Temp

-10℃ -55℃

0℃ -50℃

Cable Length

130cm

120cm

Weight55g

55g

Single-Ended Active Probe
Model

SAP1000

SAP2500

Bandwidth

1GHz

2.5GHzInput Impedance

1MΩ

1MΩ

Input Capacitance

1.2pF

1.1pF

Input Dynamic Range

±8V

±8V

Offset Range

±12V

±12V

Non-Destruct Voltage

20V

20V

Interface

SAPBus

SAPBusCable Length

130cm

130cm

Parameter2

int.siglent.com

SIGLENT Probe Data Sheet

High-Frequency Differential Active Probe
Model

SAP2500D

SAP5000D

Bandwidth (probe only)>2.5 GHz>5 GHzBandwidth (with scope)

2 GHz（SDS6204A）

4 GHz（SDS7404A）

Differential Input Capacitance

1 pF

400 fF

Differential Input Resistance

200 kΩ

20 kΩ

Single-ended Input Resistance

100 kΩ

10 kΩ

Offset Range

±8 V

±12 V

Attenuation Ratio (DC)

÷10

÷10

Offset Accuracy

< 3%

< 3%

DC Gain Accuracy< 3%< 3%Input Dynamic Range±4 V±2.5 V

Maximum Input Voltage(non-destructive)

20 V

20 V

Interface

SAPBus

SAPBusCable Length

130 cm130 cmParameter

Active Probe Adapter
Model

TPA10Parameter

Bandwidth

4GHz

Power Supplies+15V (±2%, 100 mA),
- 15V (±2%, 100 mA),
+5V (±2%, 200 mA),
- 5V (±2%, 200 mA)

Offset Range- 1.2V ~ +1.2V to probe

int.siglent.com

3

SIGLENT Probe Data Sheet

Current Probe
ModelParameter

CP4020CP4050CP4070CP4070A

Bandwidth

DC-200kHz

DC-1MHz

DC-300kHz

DC-300kHz

Rise Time

1.75μS

0.35μS

1.2μS

1.2μS

Max. effective
Value of AC

20 Arms

50 Arms

70 Arms

70 Arms

Peak-Peak Value

60 A

140 A

200 A

200 A

Range Switch

50mV/A；5mV/A

500mV/A；50mV/A

50mV/A；5mV/A

100mV/A；10mV/A±3%±20mA at 500mV/A(20mA-14A peakrange);±2%±0.4A at 50mV/A(0.4A-10A p-p range);
DC Accuracy±2%±1A at 5mV/A(1A-60A p-p range)±2%±0.4A at 50mV/A±4%±200mA at 50mV/A (0.4A-10A p-p range);(200mA-100A peakrange);±2%±1A at 5mV/A(1A-200A p-p range)±15% max at 50mV/A(100A peak-140A peakrange)

Power Supply

4±3%±50mA at 100mV/A(50mA-10A peak range);±4%±50mA at 10mV/A(500mA - 40A peakrange);±15% max at 10mV/A(40A-200A peak range)

DC 9V Adapter

Max. rated Voltage CAT III 600V,
to earth
CAT II 600V

CAT III 300V,
CAT II 600V

Conductor Size

10.3mm

10.3mmCAT III 600V,
CAT II 600V
10.3mm11mm

int.siglent.com

SIGLENT Probe Data Sheet

ModelParameter

CP6030CP6030A

CP6150CP6500

Bandwidth

DC-50MHz

DC-100MHz

DC-12MHz

DC-5MHz

Rise Time≤ 7nS≤ 3.5nS≤ 29nS≤ 70nS

Max. effective Valueof AC

30 Arms

30 Arms

150 Arms

500 Arms

Peak-Peak Value

50 A

50 A

300 A

750 A

Range

5A(1X)/30A(10X)

5A(1X)/30A(10X)

30A(10X)/150A(100X)

75A(10X)/500A(100X)

Overload Value

5A(≥5A)
30A(≥50A)

5A(≥5A)
30A(≥50A)

30A(≥30A)
150A(≥300A)

75A(≥75A)
500A(≥750A)

Current Transfer
Ratio

5A(1V/A)
30A(0.1V/A)

5A(1V/A)
30A(0.1V/A)

30A(0.1V/A)
150A(0.01V/A)

75A(0.1V/A)
500A(0.01V/A)

Measurement
Resolution

5A(1mA)
30A(10mA)

5A(1mA)
30A(10mA)

30A(10mA)
150A(100mA)

75A(10mA)
500A(100mA)

DC Accuracy

5A(±1%±1mA)
30A(±1%±10mA)

5A(±1%±1mA)
30A(±1%±10mA)

30A(±1%±10mA)
150A(±1%±100mA)

75A(±1%±10mA)
500A(±1%±100mA)

Max. rated Voltage
to earth
Conductor DiameterMax
Cable Length

300V CAT I

300V CAT III 600V CAT II

5mm

20mm

1m

1.5m

Power Supply

DC 12V/1A

BNC Length

100cm

Weightint.siglent.com

255g

555g

525g

5

SIGLENT Probe Data SheetModel

CPL5100

Parameter

6Range Level

23℃ , 60%RH, cable under test get through the test center, load resistance 1MΩ

Current Range

LH

Attenuation Accuracy

50mA~10A Peak

1A~100A Peak

Typical DC Precision

0.1V/A

0.01V/A

DC Accuracy

3%±50mA

500mA~40A Peak: 4%±50mA;
40A~100A
Peak: ±15% Maximum

Bandwidth (-3dB)

DC-600kHz

Phase Shift

DC ～ 65Hz：<1.5°

Typical DC Linearity

The typical DC linearity at H level (0.01 V/A)

Rise Time≤ 583ns

Max Operation Current

10A

Max Operation Voltage

600V

Max Floating Voltage

600V

Operating Voltage RMS

CATI 600V CATII 600V CATIII 300V

Common Mode Voltage
RMS

CATI 600V CATII 600V CATIII 300V

Typical Battery Typeand Life9 V alkaline layer-built battery/ 15 H

Low Power Indication

When battery voltage is lower than 6.5 V, battery indicator will turn red and alert

Overload Indication

When the current under test surpasses the range, the buzzer will buzz

Length of the Cable
connecting current
clamp and output box

1m

Length of double
terminal BNC Cable

1m

DC ～ 65Hz：<1°

100A

int.siglent.com

SIGLENT Probe Data Sheet

ModelParameter

SCP5030

SCP5030A

SCP5150

SCP5500

Bandwidth

DC-50MHz

DC-100MHz

DC-12MHz

DC-2MHz

Rise Time≤ 7nS≤ 3.5nS≤ 29nS≤ 175nS

Max. effective Valueof AC

30 Arms

30 Arms

150 Arms

500 Arms

Peak-Peak Value

50 A

50 A

300 A

750 A

Range

5A(1X)/30A(10X)

5A(1X)/30A(10X)

30A(10X)/150A(100X)

75A(10X)/500A(100X)

Overload Value

5A(≥5A)
30A(≥50A)

5A(≥5A)
30A(≥50A)

30A(≥30A)
150A(≥300A)

75A(≥75A)
500A(≥750A)

Current Transfer
Ratio

5A(1V/A)
30A(0.1V/A)

5A(1V/A)
30A(0.1V/A)

30A(0.1V/A)
150A(0.01V/A)

75A(0.1V/A)
500A(0.01V/A)

Measurement
Resolution

5A(1mA)
30A(10mA)

5A(1mA)
30A(10mA)

30A(10mA)
150A(100mA)

75A(10mA)
500A(100mA)

DC Accuracy

5A(±1%±1mA)
30A(±1%±10mA)

5A(±1%±1mA)
30A(±1%±10mA)

30A(±1%±10mA)
150A(±1%±100mA)

75A(±1%±10mA)
500A(±1%±100mA)

Max. rated Voltage
to earth
Conductor Diameter
Max

300V CAT I

300V CAT III 600V CAT II

5mm

20mm

Power Supply

Directly powered by the oscilloscope through SAPBUS

Weight270g

int.siglent.com

475g

7

SIGLENT Probe Data Sheet

High-Voltage Differential Active Probe
Model

DPB5150

DPB5150A

DPB5700

DPB5700A

Bandwidth（-3dB）

70MHz

100MHz70MHz

100MHzRise Time≤ 5ns≤ 3.5ns≤ 5ns≤ 3.5ns

DC Accuracy±2%±2%±2%±2%ParameterAttenuation Ratio
Max Differential Test
Voltage (DC + Peak AC)
Max Input Common
Mode voltage(voltage-to-earth Vrms)
Single-ended toInput
GroundImpedance
Two Inputs

50X/500X

100X/1000X

50X： ±150V 500X：±1500V

100X： ±700V 1000X：±7000V

600V CATIII 1000V CATII

1000V CATIII 2300V CATI

5MΩ

5MΩ

20MΩ

20MΩ

10MΩ

10MΩ

40MΩ

40MΩ

Single-ended toInput
GroundCapacitance
Two Inputs

＜ 4pF

＜ 4pF

＜ 5pF

＜ 5pF

＜ 2pF

＜ 2pF

＜ 2.5pF＜ 2.5pF

DC＞ 80dB＞ 80dB＞ 80dB＞ 80dB

100kHz＞ 60dB＞ 60dB＞ 60dB＞ 60dB

1MHz＞ 50dB

＞ 50dB＞ 50dB＞ 50dB

CMRR

Noise (Vrms)
Propagation Delay

50X: ＜50mV 500X: ＜300mV100X: ＜200mV 1000X: ＜1.2V

Probe：≈9ns

BNC Line(1m)：≈5ns

Bandwidth Limit
Differential Overvoltage
Detection Level

50X: ≥ 150V 500X: ≥ 1500V

100X: ≥ 700V 1000X: ≥ 7000V

Overload Indicator (red light)

YES

Overload Alarm

YES（Can shut up manually）

Automatic Save

YES

Offset Setting Function

YES（Set in test mode）

Terminate Load

≥100kΩ

Power Supply

USB 5V/1A Adapter

Probe Body Dimensions

195*65*28 mm

Probe Body Weight8≥ -3dB@5MHz

216g

216g

int.siglent.com

SIGLENT Probe Data Sheet

ModelParameter

DPB1300

DPB4080

Bandwidth（-3dB）

50MHz

50MHz

Rise Time

≤ 7ns

≤ 7ns

DC Accuracy

±2%

±1%

Attenuation Ratio
50X：±130V

10X： 80Vpp

500X：±1300V

100X：800Vpp

600V CATIII
1000V CATII

5kVrmsSingle-ended to
Ground5MΩ

2MΩ

Two Inputs

10MΩ

4MΩ

Single-ended to
Ground＜ 4pF＜ 2.5pF

Two Inputs＜ 2pF＜ 1.3pF

DC＞ 80dB

60Hz＞ 80dB

100kHz＞ 60dB

100Hz＞ 50dB

1MHz＞ 50dB

100kHz＞ 50dB

Max Differential TestVoltage (DC + Peak AC)
Max Input Common
Mode voltage(voltage-to-earth Vrms)
Input
Impedance
Input
Capacitance

50X/500X

CMRR

Noise (Vrms)

Propagation Delay

50X: ＜ 50mV
500X: ＜ 300mV
Probe：≈10ns
BNC Line(1m)：≈5ns

Bandwidth Limit

Null

Differential Overvoltage
Detection Level

50X: ≥ 140V
500X: ≥ 1400V

Overload Indicator (red light)

YES

Null

Terminate Load

≥ 100kΩ

1MΩ

Power Supply

DC12V/1.2A Adapter

6V DC Power

Probe Body Dimensions

145*58*24 mm

165*69*26 mm

Probe Body Weight

165g

500g

int.siglent.com

9SIGLENT Probe Data SheetHigh Voltage Probe
Model

HPB4010

Parameter

Bandwidth（-3dB）

DC-40MHz

Rise Time≤8.8nsMax. Measurement Voltage

DC: 0~10kV DC
AC: pulse ≤ 20kV peak to peak
AC: sine wave ≤ 7kV rms

Single / NoiseDC ≥ 60dB(1kHz), ≥ 50dB(1MHz)

Attenuation Ratio

1:1000

Input Impedance

100MΩ±5%Input Capacitance

3.0pF±0.5pF

Compensation Range

5pF~50pF

Cable length

2.0meter(±0.2M）

Temperature Coefficient≤ 200ppm/℃

AccuracyDC±3% (DC to 10kV)

AC±3% (1kHz/1kV/1kHz RMS)
-3dB: 0~40MHz

Operating Temperature

0~50℃

Storage Temperature-20 ~ ＋70℃

Weight / Volume

250g/Ф75×340 mm

10int.siglent.com

SIGLENT Probe Data Sheet

Logic Probe
ModelSPL2016

Parameter

SLA1016

Input Channels

1616

Input Impedance

100kΩ||18pF

100kΩ||8pF

Maximum Input Voltage±50V Peak±20V Peak

Input Dynamic Range

±20V

±10V

User defined threshold range-10V ～ 10V(10mV steps)-8V ～ 8V
(10mV steps)

TTL(1.5V)、CMOS(2.5V)、TTL(1.5V)、CMOS(2.5V)、

Threshold Selections

3.3V_LVCMOS(1.65V)、
2.5V_LVCMOS(1.25V)

3.3V_LVCMOS(1.65V)、
2.5V_LVCMOS(1.25V)

Threshold Accuracy±（3% of threshold setting +200mV）±（3% of threshold setting+150mV）Group 2：D15-D8Group 2：D15-D8Group 1：D7-D0Group 1：D7-D0

Minimum Input Voltage Swing

800mVpp

800mVpp

Maximum Input Data Rate

300 Mbps

120 Mbps

Minimum Detectable Pulse Width

3.3ns

8.3ns

Channel-to-Channel Skew

± 1 digital sample interval± 1 digital sample interval

Threshold Groupings

int.siglent.com

11SIGLENT Probe Data Sheet

Near Field ProbeModel

SRF5030T-H20

SRF5030T-H10

SRF5030T-H5

SRF5030T-E5Frequency Range

300kHz to 3 GHz

300kHz to 3 GHz

300kHz to 3 GHz

300kHz to 3 GHzResolution

20mm

10mm

5mm

5mmParameter

The SRF5030T Near Field Probe Kit includes magnetic (H) and electric (E) probes for EMC
pre-compliance testing to locate radiation sources in electronics.

Application

A near-field probe is similar to a broadband antenna, detecting radiated signals fromcomponents, PCB boards, gaps in shielding covers, etc. The use of smaller probes allows forgreater accuracy in locating the radiation area.
Other applications include: shock immunity testing, troubleshooting in RF signal chains,non-invasive testing of modulators and oscillators, measuring frequency, phase, spectralcomponents, etc. with LNAs.

12int.siglent.com

SIGLENT Probe Data Sheet

About SIGLENTSIGLENT is an international high-tech company, concentrating on R&D, sales,
production and services of electronic test & measurement instruments.

SIGLENT first began developing digital oscilloscopes independently in 2002.
After more than a decade of continuous development, SIGLENT has extended
its product line to include digital oscilloscopes, isolated handheld
oscilloscopes,

function/arbitrary

waveform

generators,

RF/MWsignalgenerators, spectrum analyzers, vector network analyzers, digital multimeters,
DC power supplies, electronic loads and other general purpose test
instrumentation. Since its first oscilloscope was launched in 2005, SIGLENT
has become the fastest growing manufacturer of digital oscilloscopes. We
firmly believe that today SIGLENT is the best value in electronic test &measurement.

Headquarters:
SIGLENT Technologies Co., Ltd
Add: Bldg No.4 & No.5, Antongda IndustrialZone, 3rd Liuxian Road, Bao'an District,Shenzhen, 518101, China

Tel: + 86 755 3688 7876
Fax: + 86 755 3359 1582
Email: sales@siglent.com
Website: int.siglent.com

North America:
SIGLENT Technologies America, Inc6557 Cochran Rd Solon, Ohio 44139
Tel: 440-398-5800
Toll Free: 877-515-5551
Fax: 440-399-1211

Email: info@siglentna.com

Follow us on
Facebook: SiglentTech

Website: www.siglentna.com

Europe:SIGLENT Technologies Germany GmbH
Add: Staetzlinger Str. 7086165 Augsburg, Germany
Tel: +49(0)-821-666 0 111 0
Fax: +49(0)-821-666 0 111 22
Email: info-eu@siglent.com
int.siglent.com

Website: www.siglenteu.com

13