SSA5000A
Spectrum Analyzer

DataSheet
EN02A

Features and Benefits

SSA5000A

Frequency Range from 9 kHz up to 13.6
GHz/ 26.5 GHz.
-165 dBm/Hz Displayed Average NoiseLevel (Typ.)-105 dBc/Hz@1 GHz,10 kHz offset SSB

General Description
The SIGLENT SSA5000A spectrum analyzers

Phase Noise (Typ.)are powerful and flexible tools for complex RF

25 MHz/40 MHz Analysis Bandwidth.

spectrum

100% POI 7.20 μs, Dynamic Range 60 dB,

andsignalanalysis.

Withacapability of real-time spectrum analysis, theanalyzer can provide multi-dimensions data

Multi-view for Density, Spectrogram anddisplays, advanced triggering to solve modern

PvT.

RF spectrum challenges, like channel power

Channel power, ACPR, OBW, Harmonic,

measurement, hopping frequency, conflict
channel, spectrum interference.

TOI measurement etc.

Applications include broadcast monitoring /Analog Modulation Analysis and Vector

evaluation,

cellular

Digital Modulation Analysis.

bluetoothsurveying,development,
maintenance.

site,

education,

IoT,

wlan

and

researchandproduction,

andBluetooth Modulation Analysis.
Pulse measurement.
Phase Noise measurement.
Noise Figure measurement.
Electromagnetic compatibility measurement.
12.1 inch Multi-Touch Screen, HDMI output.
Web Browser Remote Control on PC andMobile Terminals and File Operation.

int.siglent.com

1

SSA5000A DataSheet

Design Features
Spectrum Analyzer Mode
Various RF spectrum measurement and analysis.

Real Time Analysis Mode
Multi-view and dimensions to monitor complex signals.

2

int.siglent.com

SSA5000A DataSheet

Modulation Analysis Mode
AM/FM/PM analog modulation, and ASK/FSK/PSK/MSK/QAM digital modulation analysis.

Bluetooth signal modulation Analysis
Provide transmitter tests of Low Energy, Basic Rate and Enhanced Data Rate Bluetooth standard.

int.siglent.com

3

SSA5000A DataSheet

Noise Figure Mode
The noise figure of systems such as amplifiers and frequency converters.

Phase Noise Analysis
One-sided band phase noise test analysis, jitter analysis, etc.

4

int.siglent.com

SSA5000A DataSheet

Pulse measurement Analysis
Measure the time and frequency parameters of the RF pulses

Accessories
Utility Kit

int.siglent.com

Near Field Probe Set

USB-GPIB Adaptor

5

SSA5000A DataSheet

Definitions and Conditions
Specifications are valid under the following conditions: The instrument is within the calibration period, hasbeen stored between 0 and 50°C for at least 2 hours prior to use, and has been powered on and warmed
up for at least 40 minutes. The specifications include the measurement uncertainty, unless otherwise
noted.

Specifications: All products are guaranteed to meet published specifications when operating at room
temperature (approximately 25°C), unless otherwise noted.

Typical: Performance deemed typical implies that 80 percent of the measurement results will meet the
typical published performance with a 95th percentile confidence level at room temperature (approximately
25°C). Typical performance is not warranted and does not include measurement uncertainty.

Nominal: The expected performance or design attribute.

6

int.siglent.com

SSA5000A DataSheet

Frequency and Time Specifications
Frequency span
SSA5083A

SSA5085A

Frequency range

9 kHz ~ 13.6 GHz

9 kHz ~ 26.5 GHz

Frequency resolution

1 Hz

Range

0 Hz (Zero Span), 100 Hz to Max Frequency

Accuracy± Span / (number of sweep points - 1)

Internal Reference Source
Reference frequency

10.000000 MHz

Reference frequency accuracy± [(time since last adjustment× frequency aging rate)+ temperature stability+
initial calibration accuracy]

Standard Reference Source
Initial calibration accuracy< 1 ppm, 20 °C ~ 30 °C

Temperature stability< 1 ppm, 0 °C ~ 50 °C

Frequency aging rate

< 0.5 ppm/first year, 3.0 ppm/20 years

OCXO Precise Reference Source (Opt. 10M_OCXO_L)
Initial calibration accuracy< 0.1 ppm, 20 °C ~ 30 °C

Temperature stability< 1 ppb, 0 °C ~ 50 °C

Frequency aging rate

< 50 ppb/year

MarkerMarker resolution

Span / (number of sweep points - 1)

Marker uncertainty± [frequency indication× reference frequency uncertainty+ 10%× resolution
bandwidth+ ½ * marker resolution+ 1 Hz]

Marker type

Normal, Delta, Fixed, Reference to, Table

Marker Functions

Noise marker, N dB BW, Frequency counter

Frequency Counter resolution

0.1 Hzint.siglent.com

7

SSA5000A DataSheet

Bandwidths
Resolution bandwidth (-3dB)

1 Hz ~ 10 MHz, in 1-3-10 sequence

Resolution filter shape factor< 5:1 (60 dB:3 dB) (nom.)

RBW uncertainty< 5% (nom.)

Video bandwidth (-3dB)

1 Hz ~ 10 MHz, in 1-3-10 sequence

VBW uncertainty< 5% (nom.)

EMI bandwidth (-6dB)

200 Hz, 9 kHz, 120 kHz, 1 MHz (CISPR16 compliant)Analysis bandwidth

25 MHz, 40 MHz (opt. SSA5000-B40)

Sweep and Trigger
1 μs to 6000 s, Span = 0 Hz
Sweep time
1 ms to 4000 s, Span ≥ 100 Hz
RBW = 3k Hz ~ 10 MHz, Swept
Sweep mode
RBW = 1 Hz ~ 10 kHz, FFT
Sweep (trace) points

201 ~ 10001

Sweep rule

Single, Continuous

Trigger source

Free, Video, External, Period-200 ms ~ +500 ms, Span=0 Hz

Trigger delay
1 ms ~ +500 ms, Span ≥ 100 Hz, swept and FFTExternal trigger

5V TTL level, Rising edge/Falling edge

Time gating
Gate methods

Gated LO, Gated Video, Gated FFT

Gate length

501 μs ~ 5 s

Gate delay

2 μs ~ 25 s

8

int.siglent.com

SSA5000A DataSheet

Amplitude Accuracy and Range Specifications
Amplitude and Level
Measurement range

DANL to +10 dBm, 100 kHz ~ 1 MHz, Preamp off
DANL to +23 dBm, 1 MHz ~ 26.5 GHz, Preamp off

Reference level-170 dBm to +23 dBm, 1 dB steps

Preamplifier

20 dB (nom.), 100 kHz ~ 26.5 GHz

Input attenuation

0 ~ 50 dB, 2 dB steps

Maximum input DC voltage+/- 50 VDC
27 dBm, 3 minutes, fc ≥ 10 MHz, att > 20 dB, preamp off

Maximum average power
27+10*log(fc/1MHz) dBm, fc < 1 MHz, att ≥ 20 dB

Level Display
Logarithmic level axis

1 dB to 200 dB

Linear level axis

0 to reference level, 0% to 100%

Units of level axis

dBm, dBmV, dBμV, dBμA, Volt, Watt

Number of traces

6

Trace detectors

Positive-peak, Negative-peak, Sample, Normal, Average (Voltage/RMS/Video)

Trace functions

Clear Write, Max Hold, Min Hold, View, Blank, Average, Math

SSB Phase Noise
Offset20 °C to 30 °C, fc = 1 GHz, Normalized to 1 Hz

10 kHz-103 dBc/Hz, -105 dBc/Hz (typ.)100 kHz-103 dBc/Hz, -105 dBc/Hz (typ.)1 MHz-116 dBc/Hz, -119 dBc/Hz (typ.)int.siglent.com

9SSB Phase Noise (dBc/Hz)

SSA5000A DataSheet

fc=25 GHz

fc=500 MHz
fc=8 GHz

0.01

0.1

1

10

100

1000

10000

Frequency (kHz)Displayed Average Noise Level (DANL)SSA5083A

SSA5085A

20 °C to 30 °C, att = 0 dB, RBW = 1 Hz, sample detector, trace average > 50

Preamp off

Preamp on

10

100 kHz ~ 1 MHz-130 dBm, -143 dBm (typ.)-130 dBm, -143 dBm (typ.)1 MHz ~ 10 MHz-143 dBm, -148 dBm (typ.)-143 dBm, -148 dBm (typ.)10 MHz ~ 1.22 GHz-144 dBm, -148 dBm (typ.)-144 dBm, -148 dBm (typ.)1.22 GHz ~ 3.15 GHz-140 dBm, -144 dBm (typ.)-140 dBm, -144 dBm (typ.)3.15 GHz ~ 7.22 GHz-137 dBm, -141 dBm (typ.)-137 dBm, -141 dBm (typ.)7.22 GHz ~ 13.6 GHz-136 dBm, -140 dBm (typ.)-136 dBm, -140 dBm (typ.)13.6 GHz ~ 18.9 GHz-134 dBm, -140 dBm (typ.)18.9 GHz ~ 24.2 GHz-132 dBm, -137 dBm (typ.)24.2 GHz ~ 26.5 GHz-124 dBm, -134 dBm (typ.)100 kHz ~ 1 MHz-135 dBm, -148 dBm (typ.)-135 dBm, -148 dBm (typ.)1 MHz ~ 10 MHz-153 dBm, -165 dBm (typ.)-153 dBm, -165 dBm (typ.)10 MHz ~ 1.22 GHz-159 dBm, -163 dBm (typ.)-159 dBm, -163 dBm (typ.)1.22 GHz ~ 3.15 GHz-158 dBm, -162 dBm (typ.)-158 dBm, -162 dBm (typ.)3.15 GHz ~ 7.22 GHz-154 dBm, -158 dBm (typ.)-154 dBm, -158 dBm (typ.)7.22 GHz ~ 13.6 GHz-154 dBm, -158 dBm (typ.)-154 dBm, -158 dBm (typ.)13.6 GHz ~ 18.9 GHz-151 dBm, -155 dBm (typ.)18.9 GHz ~ 24.2 GHz-148 dBm, -152 dBm (typ.)24.2 GHz ~ 26.5 GHz-142 dBm, -149 dBm (typ.)int.siglent.com

SSA5000A DataSheet

Error and Accuracy
Resolution bandwidth
switching uncertainty

Logarithmic resolution, relative to RBW = 10 kHz± 0.2 dB (nom.)
20 °C to 30 °C, preamp off, relative to att = 20 dB

Input attenuation switching
uncertainty

1 MHz ~ 7.22 GHz

0.5 dB

7.22 GHz ~ 26.5 GHz

0.7 dB

20 °C to 30 °C, 30% to 70% relative humidity, att = 20 dB, preamp off
Frequency Response

10 MHz ~ 7.5 GHz± 1.5 dB

relative to 50 MHz

7.5 GHz ~ 13.6 GHz± 2.0 dB

13.6 GHz ~ 26.5 GHz± 2.5 dB

20 °C to 30 °C, input signal level = -50 ~ -10 dBm, RBW = VBW = 30 kHz, att = 20 dB,
peak detector, 95% reliability
Absolute amplitude accuracy± 0.4 dB, fc = 50 MHz± 0.4 dB + Frequency Response, Preamp off± 0.5 dB + Frequency Response, Preamp on
Att = 10 dB, fc ≥ 10 MHz
10 MHz ~ 13.6 GHz

1.6 (nom.)

13.6 GHz ~ 24.2 GHz

1.9 (nom.)

24.2 GHz ~ 26.5 GHz

2.1 (nom.)

RF input VSWR

Distortion and Spurious Responses
20 °C to 30 °C, fc ≥ 50 MHz, input signal level = -20 dBm, att = 0 dB, preamp off
Second harmonic
distortion (SHI)

10 MHz ~ 7.22 GHz

42 dBm (nom.)

7.22 GHz ~ 13.25 GHz

54 dBm (nom.)

20 °C to 30 °C, fc ≥ 50 MHz, input signal two -20 dBm , tones spaced by 100 kHz,
att = 0 dB, preamp off
Third-order intercept (TOI)

1dB gain compression

50 MHz ~ 7.22 GHz

11 dBm , 15 dBm (nom.)

7.22 GHz ~ 26.5 GHz

10 dBm , 14 dBm (nom.)

20 °C to 30 °C, fc ≥ 50 MHz, input signal two tones spaced larger than 20 MHz,
att = 0 dB, preamp off> 5 dBm (nom.)
20 °C to 30 °C, input terminated 50 Ω, att = 0 dB

Residual response< -90 dBm
20 °C to 30 °C, mixer level -30 dBm
Input related spurious< -65 dBc

int.siglent.com

11

SSA5000A DataSheet

Advanced Measurement Kit (SSA5000-AMK)
Power Measurement
CHP, Channel Power

Channel Power, Power Spectral Density

ACPR, Adjacent Channel Power Ratio

Main CH Power, Left channel power, Right channel power

OBW, Occupied Bandwidth

Occupied Bandwidth, Transmit Frequency Error

T-Power, Time Domain Power

Zero Span Integrated Power

CNR, Carrier Noise Ratio

C/N, Noise Power

Non-Linear Measurement
Harmonic measurement

Max Harmonic number 10

TOI, Third-Order Intercept

Measure the third-order products from two tones

Spectrum Monitor Measurement
Spectrogram

IQ Data Acquisition（Option SSA5000-IQA）Time record length (IQ pairs)
Data packing

I=Q=4 Byte

Storage depth

1 GByteLength(IQ sample pairs)

125 MByte (1GB/8B)

Length (time units)

Samples / (Span x 1.25), nominal

12

int.siglent.com

SSA5000A DataSheet

Real-Time Spectrum Analysis (Option SSA5000A-RTA1)
Frequency and Time
25 MHz (Default)
Real-Time Bandwidth
40 MHz (Option SSA5000A-B40)
100% POI Minimum
Signal Duration

Full Span, Kaiser Window, Frequency Mask Triggering
7.20 μs
Density

30 ms ~ 50 s

3D+Spectrogram

30 ms ~ 50 s

Spectrogram

100 μs ~ 50 s

PvT+Spectrum

100 μs ~ 50 s

Measurement view

MAX Sample rate

51.2 MHzFFT per second

150 000 (40 MHz analysis BW)

Marker8

Span min

5 kHz

Window

Kaiser (Default), Hanning, Flattop, Gaussian, Blackman-Harris, Rectangular
Any SPAN, six RBW for every window (only one for Rectangular), default min RBW.
Typical RBW for Kaiser:

RBW

Spectrogram / PvTMaximum stored

int.siglent.com

Span

RBW min

RBW MAX

40 MHz100.43 kHz

3.3142 MHz

20 MHz50.21 kHz1.657 MHz

10 MHz25.11 kHz828.55 kHz

1 MHz

2.51 kHz82.85 kHz

100 kHz251 Hz8.285 kHz

50 000 (Loop store)

13

SSA5000A DataSheet

Amplitude Accuracy and Range
Detector+Peak, -Peak, Sample, Average

Trace

3

Spectrum Density Display

0 ~ 100% (resolution 0.1%)

Dynamic range forSpectrogram

200 dB

Amplitude

TriggerFrequency Mask Trigger(FMT)

Colour Mode

Flatness< 0.4 dB

Resolution

0.01 dB

Dynamic range< 60 dB

Free Run, PvT, External
SourceTraces

Type

Greater Than, Less Than, Outside Mask, Inside Mask

Actions

Stop, Beep

Warm (Default), Cool, Gray

Different RBW and span, 100% POI (μs)
Analysis BW

RBW1

RBW2

RBW3

RBW4

RBW5

RBW6

40 MHz

26.56

16.56

11.56

9.06

7.81

7.20

20 MHz

46.56

26.56

16.56

11.56

9.06

7.81

10 MHz

86.56

46.56

26.56

16.56

11.56

9.06

1 MHz

806.56406.56206.56106.56

56.56

31.56

Different window length for RBW
Length\Type

1024

512

256

128

64

32

Kaiser (Beta=12)398.2849

198.9478

99.279349.4450

24.5279

12.0693

Hanning

533.4785

266.4785

132.9785

66.2285

32.8535

16.1660

Flattop212.2447

106.0182

52.9050

26.3483

13.0700

6.4309Gaussian (alpha=3.5)

404.8707

202.2399

100.9244

50.2666

24.9376

12.2729

Blackman-Harris399.2401

199.4250

99.5174

49.5636

24.5868

12.0983

Rectangular801

400.5000

200.2500

100.1250

50.062525.0313

14

int.siglent.com

SSA5000A DataSheet

Pulse Measurement (Option SSA5000-PU)
Measurement
Sweep Points

201 ~ 10001

Trace Function

Clear/Write, Max Hold, Min Hold, Average

Trace Number

1
Positive Peak,
Negative Peak

Detector

Sample,Average (Video/RMS/V)
NormalTrace Average (all the points in the trace),
Positive Peak, Negative Peak,Power Measurement

Wave Average (all points within all complete periods that are available on the trace),
Pulse Average (the points in the high state of the pulse),
Peak-Wave Average
Duration,
Pulse Center, Period,
Off Time (positive pulse only)
Duty Fact,

Pulse Charateristics

Frequency,
Rising Time, Falling Time,
Pre-Overshoot (first pulse only),
Pre-Undershoot (first pulse only),
Post-Overshoot (first pulse only),
Post-Undershoot (first pulse only)

int.siglent.com

15

SSA5000A DataSheet

Phase Noise Measurement (Option SSA5000-PN)
Measurement
SSA5083A

SSA5085A

Frequency range

100 kHz ~ 13.6 GHz

100 kHz ~ 26.5 GHz

Minimum offset frequency

100Hz

Maximum offset frequency

13.6 GHz-Fc

Sweep Points

601 ~ 10001

26.5 GHz-Fc

Measurement function
View

Log Plot, Decade Table, Spurious Table

Trace

RAW, Smoothed, Reference, Blank

Marker function

Integrated noise , Residual FM , Averaged noise density

PatternPhase Noise, DANL Floor

Instrument phase-noise performance (Center Freq = 1GHz)
Freq

20℃ ~ 30℃ , Center Freq = 1 GHz , Normalized to 1 Hz

1 kHz

-98 dBc/Hz

10 kHz-103 dBc/Hz , -105 dBc/Hz (representative value)

100 kHz-103 dBc/Hz , -105 dBc/Hz (representative value)

1 MHz-116 dBc/Hz , -119 dBc/Hz (representative value)

10 MHz-128 dBc/Hz

16

int.siglent.com

SSA5000A DataSheet

Noise Figure Measurement (Option SSA5000-NF)
Measurement
Support device type

Amplifier, upconverter, downconverter
Noise Figure,
Gain,
Noise Factor,

measurement result

T effective,
Y factor,
P hot,
P coldRBW range

1 Hz ~ 3 MHz

Line loss compensation

Input line loss, and output line loss

The results show the pattern

Graph, Table, Meter

Noise source interface

USB-A Host to BNC (+28V)

int.siglent.com

17

SSA5000A DataSheet

EMI Measurement (Option SSA5000-EMI)
Measurement
Measurement View

Frequency scan, Meter, Signal list

Pre-compliance Sequence

Scan, Search, Meas

EMI filter RBW (-6dB)

200 Hz, 9 kHz, 120 kHz, 1 MHz (following CISPR 16-1-1)

RBW uncertainty< 5% (nom.)

Detector

Peak, Voltage Average, Quasi-Peak (following CISPR 16-1-1)

Dwell time

0 us ~ 10 s

RBW/Steps

0.1, 0.3, 0.5, 1, 2, 3

Corrections

4

Limit and Trace

3

Limit Standards

EN550xx, GB9254, FCC Part15, User defined

Attenuator

0-50 dB

ReportSignal List

Frequency scale

Linear, Logarithmic

18

int.siglent.com

SSA5000A DataSheet

Analog Modulation Analysis (Option SSA5000-AMA)
Common Parameter
SSA5083A

SSA5085A

Carrier Frequency Range

2 MHz ~ 13.6 GHz

2 MHz ~ 26.5 GHz

Carrier Power Accuracy± 2 dB (nom.)

Carrier Power Range-30 dBm to +20 dBm (nom.)AM
Modulation rate range

20 Hz to 100 kHz
1 Hz (nom.)

Modulation rate < 1 kHz< 0.1% modulation rate (nom.)

Modulation rate ≥ 1 kHz

Accuracy
Modulation depth range

5% to 95%

Accuracy±4% (nom.)FM
Modulation rate range

20 Hz to 200 kHz
1 Hz (nom.)

Modulation rate < 1 kHz< 0.1% modulation rate (nom.)

Modulation rate ≥ 1 kHz

AccuracyFrequency deviation

1 kHz to 400 kHz

Accuracy±4% (nom.)PM
Modulation rate range

50 Hz ~ 50 kHz
1 Hz (nom.)

Modulation rate < 1 kHz< 0.1% modulation rate (nom.)

Modulation rate ≥ 1 kHz

Accuracy
Phase deviation

0.2 ~ 100 rad

Accuracy±4% (nom.)int.siglent.com

19

SSA5000A DataSheet

Digital Modulation Analysis (Option SSA5000-DMA)
Common Parameter
SSA5083A

SSA5085A

Frequency Range

2 MHz ~ 13.6 GHz

2 MHz ~ 26.5 GHz

Carrier Power Accuracy± 2 dB (nom.)

Carrier Power Range-30 dBm to +20 dBm (nom.)

Measurement
ASK: 2ASK;
FSK: 2FSK, 4FSK, 8FSK, 16FSK;
Modulation TypeMSK: GMSK;PSK: BPSK, QPSK, OQPSK, 8PSK;DPSK: DBPSK, DQPSK, D8PSK, π/4 -DQPSK, π/8 -D8PSK;
QAM: 16, 32, 64, 128, 256

Meas Length

16 to 4096

Points / Symbol

4, 6, 8, 10, 12, 14, 16

Symbol Rate

1 ksps to 32 Msps, Symbol Rate* Points / Symbol ≤ 150 Msps

Trigger Holdoff

500 msBurst

Burst power sync, BERT

FilterMeas / Ref Filter

Nyquist, Squrt Nyquist, Gauss, Half Sine, Rectangular

Length2 to 128

Alpha/BT

Alpha 0.01 ~ 1, BT 0.01 ~ 10

Trace
IQ Meas Time, IQ Meas Spectrum, IQ Ref Time, IQ Ref Spectrum,
Trace Data

Time, Spectrum, IQ Mag Err, IQ Phase Err,
Symbol Error Chart, Err Vector Time, Err Vector Spectrum,

Trace Formats

Log mag, Lin mag, Real, Imag, I-Q, Constellation, I-eye, Q-eye, Wrap Phase,
Unwrap Phase, Trellis eye

Symbol Error Chart
EVM (rms EVM, peak EVM), Magnitude error,
PSK/ DPSK/ MSK/ QAM

Phase error, IQ offset, Carrier offset, SNR Quadrature error,
Gain imbalance (not support for MSK)ASKASK Error, ASK depth, carrier offset

FSK

FSK Error, Magnitude error, FSK deviation, carrier offset

20

int.siglent.com

SSA5000A DataSheet

Bluetooth Analysis (Option SSA5000-BT)
Measurement
Bluetooth type

Low Energy

Basic Rate

Enhanced Data Rate

Support standards

RFPHY.TS.p17

RF.TS.p32

RF.TS.p32

Database type

Load type

Reference Packet(1M,2M,Coded)

2-DH1/2-DH3/2-DH5/
DH1/DH3/DH5/HV3

3-DH1/3-DH3/3-DH5/
2-EV3/2-EV5/3-EV3/3-EV5

PN9,0x00,0XFF,

PN9,0x00,0XFF,

PN9,0x00,0XFF,

0x0F,0x55

0x0F,0x55

0x0F,0x55
Freq Offset ωi

Peak Power
Average Power
Freq Drift
Max Drift Rate
measurement result(subject to the load)

ΔF1 Avg/Max
ΔF2 Avg/Max
ΔF1/ΔF2 Ratio
Freq Offset
Init Freq Drift

Freq Offset ω0Peak Powerωi +ω0

Average Powerrms DEVM

Freq Driftpeak DEVM

Max Dritf Rate

99% DEVM

ΔF1 Avg/Max

GFSK Average Power

ΔF2 Avg/Max

DPSK Average Power

ΔF1/ΔF2 Ratio

Rel Average Power

ICFT

BER
Bit Errors
Guard IntervalFrequency range

2.4020 GHz ~ 2.4835 GHz

Input signal level-40 dBm ~ +20 dBm

Trigger

Free Run, Video, External, Periodic

int.siglent.com

21

SSA5000A DataSheet

Inputs and Outputs
Front Panel
RF input

2.92 mm type Male, 50 Ω (nom.)

USB host

4 USB-A plug, V2.0

Rear PanelUSB device

USB-B plug, V2.0

LAN

10/100 Base, RJ-45

Video out

HDMI

Audio out

3.5 mm JackIF output

Center frequency

120 MHz, 50 Ω, BNC-type Female

Conversion gain

-13 to +5 dB (nom.) plus frequency response

Bandwidth

40 MHz (nom.)(SSA5000-IFO)
10 MHz reference output

10 MHz, > 0 dBm, BNC-type Female, 50 Ω (nom.)

10 MHz reference input

10 MHz, -5 to +10 dBm, BNC-type Female, 50 Ω (nom.)

External trigger output

3.3V TTL level, 20 mA max, BNC-type Female

External trigger input

5V TTL level, 1 kΩ, BNC-type Female

Remote Control
Communication Interface

LAN, USB Device, USB Host (USB-GPIB adaptor)
SCPI / Labview / IVI based on USB-TMC / VXI-11 / Socket / Telnet;

Remote Control Capability

NI-MAX;Web Browser (HTML 5 Supported);22

int.siglent.com

SSA5000A DataSheet

General Specification
Structure
Dimensions

378 mm x 284 mm x 126 mm (W×H×D)

Weight

Net: 7.40 kg (16.3 lb)

Display

TFT LCD, 1200 × 800, 12.1 inch capacitive multi-touch screen

Storage

Internal (eMMC) 4 GB, external (USB storage device) 32 GB

Working Environment
Power Source

Temperature

Humidity

AC voltage range: 100-240 V, 50/60 Hz or 100-120 V, 400 Hz;
Power consumption: 80 W (MAX)Working temperature: 0 °C to 50 °C,
Storage temperature: -20 °C to 70 °C
0 °C to 30 °C, ≤ 95% Relative humidity
30 °C to 50 °C, ≤ 75% Relative humidity

AltitudeOperating: less than 3000 m

Calibration cycle

1 yearElectromagnetic Compatibility
EN 61326-1: 2013 /Class A (The active input power of the EUT is less than 75 W.

EN 61000-3-2: 2014

According to EN 61000-3-2, no limits are necessary.)

EN 61000-3-3: 2013

Plt: 0.65, Pst: 1.00 , dmax: 4.00 % , dc: 3.00 %;dt Lim: 3.30 % dt > Lim: 500ms

IEC 61000-4-2: 2008

AD ±8.0 kV , CD ±4.0 kV

IEC 61000-4-3: 2006 +80 MHz to 1000 MHz: 10V/m, 1.4 GHz to 2.0 GHz:3 V/m,

A1: 2007 + A2: 2010

2.0 GHz to 2.7 GHz:1V/m

IEC 61000-4-4: 2004

AC Line: ±2.00 kV

IEC 61000-4-5: 2005

Line to Line: 1.0 kV, Line to Earth: 2.0 kV

IEC 61000-4-6: 2008

0.15-80 MHz:3 V 1 KHz 80% AM

IEC 61000-4-8: 200930 A/m, 50/60 Hz

IEC 61000-4-11: 2004

Voltage Dips:0%/0.5P; 40%/10P; 70%/25P;
Short Interruptions Test Level % UT: 0%/250P

SafetyIEC 61010-1:2010/EN 61010-1:2010
CAN/CSA-C22.2 No.61010-1:2012, UL 61010-1:2012,
CAN/CSA-C22.2 No.61010-2-30:2012, UL 61010-2-30:2012
RoHS
2011/65/EU

int.siglent.com

23

SSA5000A DataSheet

Ordering Information
Product

Description

Order Number

Spectrum Analyzer, 9 kHz ~ 13.6 GHz

SSA5083A

Spectrum Analyzer, 9 kHz ~ 26.5 GHz

SSA5085A

Product CodeStandard

Options

Accessories

Quick Start, USB Cable, Power Cord, Wireless Mouse,
2.92F-2.92F-40A
SSA5083A upgrade to SSA5085A

SSA5000-F5

Pre-Amplifier, 9 kHz ~ 13.6 GHz

SSA5000-P3

Pre-Amplifier, 9 kHz ~ 26.5 GHz

SSA5000-P5

40 MHz analysis bandwidth

SSA5000-B40

Real-Time Spectrum Analysis

SSA5000-RTA1

Advanced Measurement Kit

SSA5000-AMK

IQ Data Acquisition

SSA5000-IQA

Pulse Measurement

SSA5000-PU

Phase Noise Measurement

SSA5000-PN

Noise Figure Measurement

SSA5000-NF

Analog Modulation Analysis

SSA5000-AMA

Digital Modulation Analysis

SSA5000-DMA

Bluetooth Analysis

SSA5000-BT

EMI Measurement

SSA5000-EMI

IF Output

SSA5000-IFO

OCXO Precise Reference source, Factory installed

10M_OCXO_L

2.92mm(F)-2.92mm(F) adaptor, DC ~ 40 GHz

2.92F-2.92F-40A

N(M)-N(M) cable, DC ~ 18 GHz, 1000 mm

N-N-18L

N(M)-SMA(M) cable, 18 GHz, 1000 mm

N-SMA-18L

SMA(M)-SMA(M) cable, 18 GHz, 1000 mm

SMA-SMA-18L

SMA(M)-SMA(M) cable, 26.5 GHz, 1000 mm

SMA-SMA-26L

SMA(F)-SMA(M) cable, 26.5 GHz, 1000 mm

SMAF-SMA-26L

USB-GPIB Adaptor

USB-GPIB

300 kHz~3 GHz Near Field Probe Kit:
3 H-probes (20/10/5 mm), 1 E-probe (5 mm)

24

SRF5030T

int.siglent.com

About SIGLENTSIGLENT is an international high-tech company, concentrating on R&D,
sales, production and services of electronic test & measurement instruments.
SIGLENT first began developing digital oscilloscopes independently in 2002.
After more than a decade of continuous development, SIGLENT has
extended its product line to include digital oscilloscopes, isolated handheld
oscilloscopes, function/arbitrary waveform generators, RF/MW signal
generators, spectrum analyzers, vector network analyzers, digital
multimeters, DC power supplies, electronic loads and other general purpose
test instrumentation. Since its first oscilloscope was launched in 2005,
SIGLENT has become the fastest growing manufacturer of digital
oscilloscopes. We firmly believe that today SIGLENT is the best value inelectronic test & measurement.

Headquarters:
SIGLENT Technologies Co., Ltd
Add: Bldg No.4 & No.5, Antongda Industrial
Zone, 3rd Liuxian Road, Bao'an District,
Shenzhen, 518101, China
Tel: + 86 755 3688 7876
Fax: + 86 755 3359 1582
Email: sales@siglent.com
Website: int.siglent.com

North America:
SIGLENT Technologies America, Inc
6557 Cochran Rd Solon, Ohio 44139
Tel: 440-398-5800
Toll Free: 877-515-5551
Fax: 440-399-1211
Email: info@siglentna.com
Website: www.siglentna.com

Europe:SIGLENT Technologies Germany GmbH
Add: Staetzlinger Str. 70
86165 Augsburg, Germany
Tel: +49(0)-821-666 0 111 0
Fax: +49(0)-821-666 0 111 22
Email: info-eu@siglent.com
Website: www.siglenteu.com

Follow us on
Facebook: SiglentTech