SDG6000X
SeriesDataSheet-2022.07

Pulse/Arbitrary
Waveform Generator

SDG6000X Pulse/Arbitrary Waveform Generator

Key Features
Dual-Channel, 500 MHz maximum bandwidth, 20 Vpp

SDG6052X

maximum output amplitude, high fidelity output with 80

SDG6032X

High-performance sampling system with 2.4 GSa/s

SDG6022X

Multi-function signal generator, meeting requirements indB dynamic range
sampling rate and 16-bit vertical resolution
wide range

Overview
SIGLENT’s SDG6000X is a series of dual-channel Pulse/
Arbitrary Waveform Generators that feature up to 500 MHz
bandwidth, a maximum sample rate of 2.4 GSa/s and16-bit vertical resolution. They also include proprietary
TrueArb & EasyPulse technology that help to solve theweaknesses inherent in traditional DDS generators when

Continuous
Wave
Generator

Up to 500 MHz sine wave, supporting
sweep and user-defined harmonics.
Low cost replacement of RF signal
generators below 500 MHz

Pulse
Generator

Up to 150 MHz Pulse, with finely adjustable
width, rising edge and falling edge; 3.3 ns
minimum width and 1 ns minimum edge atfull frequency range

Function
Arbitrary
Waveform
Generator

Basic Function/Arbitrary Waveform
Generator and complex signals generating
capability including modulation, sweep,
burst and waveform combination.

IQ Signal
Generator(optional)

Base Band and IF IQ signals supporting
basic modulation and an arbitrary symbol
rate between 250 Symb/s ~ 37.5 MSymb/s

Noise
Generator

Up to 500 MHz bandwidth White Gaussian
Noise with adjustable bandwidth

PRBS
Generator

Up to 300 Mbps PRBS3 ~ PRBS32 with fine
bit rate and edge adjustments

generating arbitrary, square and pulse waveforms. In
addition, the SDG6000X is a multi-function device which
can generate Noise, IQ signals and PRBS patterns. Thesefeatures enable the SDG6000X to provide a variety ofhigh fidelity and low jitter signals, meeting the growing
requirements of complex and intensive applications.

Sweep and Burst function
Harmonics function
Waveform Combining function
Channel Coupling, Copy and Tracking function
196 built-in arbitrary waveforms
High precision Frequency Counter
Standard interfaces include: USB Host, USB Device（USBTMC）, LAN（VXI-11, Socket, Telnet）. Optional

Interface: GPIB

4.3” touch screen display for easier operation
Model

SDG6022X

SDG6032X

SDG6052X

Bandwidth

200 MHz

350 MHz

500 MHz

Number of channels

2

Sampling rate

2.4 GSa/s (2X Interpolation)

Vertical resolution

16 bitArbitrary waveform length

2 ~ 20 Mpts

Display

4.3” touch screen display, 480 x 272 x RGB

Interface

Standard: USB Host, USB Device, LAN
Optional: GPIB (USB-GPIB adaptor)
WWW.SIGLENT.COM

SDG6000X Pulse/Arbitrary Waveform Generator

Characteristics
Continuous Wave

Up to 500MHz continuous sine wave.

Pulse
Adjustable Pulse Width
The pulse width can be fine-tuned to the minimum of 3.3ns with an
adjustment step as small as 100 ps, at any frequency.

Adjustable Edge
The rise/fall times can be set independently to the minimum of 1ns
at any frequency with a minimum adjustment step as small as 100
ps.

Low Jitter
When a Square/Pulse waveform is generated by traditional DDS,
there can be additional jitter if the sampling rate is not an integerrelated multiple of the output frequency. EasyPulse technology
successfully overcomes this weakness in DDS designs and helps to
produce low jitter Square/Pulse waveforms.
WWW.SIGLENT.COM

SDG6000X Pulse/Arbitrary Waveform Generator

Arbitrary Waveform
Traditional DDS designs can lead to additional jitter and distortion when sourcing arbitrary waveforms. The SIGLENT TrueArb design minimizes jitter and distortion to
help deliver high fidelity arbitrary waveforms.

Point by Point Output

Low Jitter

TrueArb generates arbitrary waveforms point-by-point. It never skips

As with EasyPulse, TrueArb effectively overcomes the clock jitter that can

any point so that it can reconstruct all the details of the waveform, as

effect traditional DDS generators.

defined. Two interpolation modes are available: linear and zero-orderhold.

Arbitrary Waveform Software EasyWaveX
EasyWaveX is an arbitrary waveform software platform that supports
waveform creation and editing. It features manual drawing, as-wellas line, equation, and coordinate editing modes. It is also a convenient
way for users to edit their own arbitrary waveforms.

Noise

Gaussian noise with bandwidth up to 500 MHz. The repetition period is

Wideband Gaussian noise can be easily added to other waveforms tomore than 100 years, and the bandwidth is adjustable.

simulate real-world scenarios in which the signal contains a large degreeof noise.

WWW.SIGLENT.COM

SDG6000X Pulse/Arbitrary Waveform Generator

IQ (optional)

The SDG6000X supports popular modulation types such as ASK, FSK, PSK, and QAM. Proprietary
resampling technology provides excellent EVM performance at arbitrary symbol rates between 250
Symb/s ~ 37.5 MSymb/s. Built-in digital quadrature modulator provides the possibility to generate IQ
signals from baseband to 500 MHz intermediate frequency.

IQ waveforms can be generated by the PC software EasyIQ.

WWW.SIGLENT.COM

SDG6000X Pulse/Arbitrary Waveform Generator

PRBS

PRBS3 ~ PRBS32 with finely adjustable 10-6 bps ~ 300 Mbps bit rate and1 ns ~ 1us edge.

Preset common logic levels such as TTL, LVCMOS, LVPECL and LVDS. An
added differential mode provides an easy way to generate differential
signals using the both channels.

High Fidelity Output with 80dB Dynamic Range

Large Signals at High Frequencies

Small Signals

Dual-channel, 20 Vpp amplitude sine wave guaranteed at up to 40 MHz.

Low noise floor, improves signal-to-noise ratio.

WWW.SIGLENT.COM

SDG6000X Pulse/Arbitrary Waveform Generator

Complex Signals Generation

Modulation

Sweep and Burst

Plenty of modulation types, such as AM, FM, PM, FSK, ASK, PSK, DSB-

Sweep modes include “Linear” and “Log”. Burst modes includes “N

AM, PWM are supported. The modulation source can be configured as

cycle” and “Gated”. Both Sweep and Burst can be triggered by “Internal”“Internal” or “External”.

, “External” or “Manual” source.

Waveform Combining
The waveform combining function
superimposes CH1 and CH2 waveforms
internally and provides the combined
waveform to a user-selected output. Easily
combine basic waveforms, random noise,
modulation signals, sweep signals, burst
signals, EasyPulse waveforms and TrueArb
waveforms

Harmonics Function
Harmonics function gives you the ability to add
higher-order elements to your signal.

Two Dual-channel Operation Mode

Frequency Counter"Phase-Locked" mode automatically aligns the phases of each output.

8-digit hardware frequency counter with statistics function and input

While "Independent" mode permit the two channels to be used as two

range of 0.1 Hz ~ 400 MHz.

independent generators. Independent mode also smoothes parameter(frequency, amplitude) changes made to an active channel.
WWW.SIGLENT.COM

SDG6000X Pulse/Arbitrary Waveform Generator

Specifications
All specifications apply to both channels. Unless otherwise stated, all specifications are not guaranteed unless the following conditions are met:
• The generator is within the valid calibration period• The generator has been working continuously for at least 30 minutes at a specified temperature (18 ℃ ~ 28 ℃ )Frequency
Parameter

Min.

Resolution

1μ

Typ.

Max.

Unit

Condition & Note

Hz

-1

+1

ppm

25℃

-2

+2

ppm

0~40℃

1 -year aging

-1

+1

ppm

25℃

10-year aging

-3.5

+3.5

ppm

25℃

Max.

Unit

Condition & Note

1μ

500M

Hz

SDG6052X

1μ

350M

Hz

SDG6032X

1μ

200M

Hz

SDG6022X-65

dBc

0 dBm, 0~1 MHz ( included )

-60

dBc

0 dBm, 1~60 MHz ( included )

-50

dBc

0 dBm, 60~100 MHz ( included )Initial accuracyst

Sine
Parameter

Frequency

Min.

Harmonic distortion

-40

dBc

0 dBm, 100~200 MHz ( included )

-30

dBc

0 dBm, 200~300 MHz ( included )

-28

dBc

0 dBm, above 300 MHz

0.075

%

0 dBm, 10 Hz ~ 20 kHz

-60

dBc

0 dBm, ≤350 MHz

-55

dBc

0 dBm, >350 MHz

2m

20

Vpp

≤ 40 MHz, HiZ load

2m

10

Vpp

40 MHz ~ 120 MHz ( included ), HiZ load

2m

5

Vpp

120 MHz ~ 160 MHz ( included ), HiZ load

2m

3

Vpp

160 MHz ~ 350 MHz ( included ), HiZ load

2m

1.28

Vpp

above 350MHz, HiZ load

Total Harmonic
Distortion
Non-harmonic spurious

Output Range (Note)

Harmonics Order
Type

Typ.

10
Even, Odd, All

Note : The specification will be divided by 2 while applied to a 50Ω load.

WWW.SIGLENT.COM

SDG6000X Pulse/Arbitrary Waveform Generator

Pulse
Parameter
Frequency

Pulse Width
Pulse width resolution

Min.

Typ.

Max.

Unit

Condition & Note

1μ

150 M

Hz

SDG6052X, SDG6032X

1μ

80 M

Hz

SDG6022X

ns

SDG6052X, SDG6032X

3.4

ns

SDG6022X

100

ps

3.3

Pulse width accuracy±(0.01%+0.3ns)
1n

75

s

SDG6052X, SDG6032X
10% ~ 90%, 100 ps resolution

2n

75

s

SDG6022X
10% ~ 90%, 100 ps resolution

1n

75

s

SDG6052X, SDG6032X
90% ~ 10%, 100 ps resolution

2n

75

s

SDG6022X
90% ~ 10%, 100 ps resolution

Rise time(specified range)

2n

75

s

Fall time(specified range)

2n

75

s

Rise/fall times resolution

100

Rise time( setting range )Fall time(setting range )psOvershootDuty cycle

0.001

Duty cycle resolution

0.001

3

%

100 kHz, 1 Vpp, 50Ω load , 2 ns edge

99.999%Limited by frequency setting%

J i tt e r ( r m s ) c yc l e t o
cycle

Output Range (Note)

10% ~ 90%, 100 ps resolution. Overshoot, jitter, output range
and pulse width accuracy specifications are only guaranteed in
specified rise/fall times range

100

ps

1 Vpp, 50Ω load

2m

20

Vpp

≤ 20 MHz, HiZ load , 2ns edge , ≥ 10 ns width

2m

10

Vpp

20 MHz ~ 120 MHz ( included ), HiZ load , 2ns edge , ≥ 10 ns
width

2m

5

Vpp

Above 120 MHz , HiZ load , 2ns edge , ≥ 10 ns width

Note : The specification will be divided by 2 while applied to a 50Ω load.

SquareParameter
Frequency

Min.

Max.

Unit

Condition & Note

1μ

120M

Hz

SDG6052X, SDG6032X

1μ

80M

Hz

SDG6022X

2.4

ns

10% ~ 90%, 1 Vpp, 50Ω load

3%

100 kHz, 1 Vpp, 50Ω load

90%

Limited by frequency setting

100

ps

1 Vpp, 50Ω load

2m

20

Vpp

≤ 20 MHz, HiZ load

2m

10

Vpp

Above 20 MHz , HiZ load

Condition & Note

Rise /fall times

Typ.

2

OvershootDuty cycle10

J i tt e r ( r m s ) c yc l e t o
cycle
Output Range (Note)

Note : The specification will be divided by 2 while applied to a 50Ω load.

Ramp
Parameter

Min.

Max.

Unit

Frequency

1μ

5M

Hz

Symmetry

0

100

%

1

%

20

Vpp

Linearity
Output Range (Note)

2m

Typ.

Note : The specification will be divided by 2 while applied to a 50Ω load.
WWW.SIGLENT.COM

Percentage of peak output, 1 kHz, 1 Vpp, 50% symmetrySDG6000X Pulse/Arbitrary Waveform Generator

Noise
Parameter

Min.

Bandwidth (-3dB)

Typ.

Max.

Unit

Condition & Note

500

MHz

SDG6052X

350

MHz

SDG6032X

200

MHz

SDG6022X

Bandwidth setting range

1m

BW

Hz

BW is the max. frequency

Output Range (Note)

2m

1.084

Vrms

Mean = 0
Bandwidth limit = OFF

Condition & Note

Note : The specification will be divided by 2 while applied to a 50Ω load.

Arbitrary Wave
Parameter

Min.

Max.

Unit

Frequency setting range

1μ

50M

Hz

Waveform length

2

20M

pts

1u

300M

Sa/s

TrueArb mode

Sa/s

DDS mode

Sampling rate

Typ.

1.2G

Vertical resolution

16

bit

Rise/fall times

2.6

ns

10% ~ 90%, 1Vpp step signal , DDS mode

100

ps

1 Vpp, 50Ω load , TrueArb mode

2m

20

Vpp

≤ 20 MHz, HiZ load

2m

10

Vpp

Above 20 MHz , HiZ load

Max.

Unit

Condition & Note

-10

10

V

HiZ load-55V50Ω loadJ i tt e r ( r m s ) c yc l e t o
cycle
Output Range (Note)

Note : The specification will be divided by 2 while applied to a 50Ω load.

DC
Parameter
Output Range

Min.

Typ.

Accuracy±(1%+2mV)

HiZ load

IQ (optional)
Parameter

Min.

Symbol rate

250

Vertical resolution

Typ.

Max.

Unit

Condition & Note

37.5M

Symb/sLimited by the oversampling factor

16

bit

Modulation type2ASK, 4ASK, 8ASK, BPSK, QPSK, 8PSK, DBPSK, DQPSK, D8PSK,
8QAM, 16QAM, 32QAM, 64QAM, 128QAM, 256QAM, 2FSK, 4FSK,
8FSK, 16FSK, MSK, MultiTone, custom

Supported by EasyIQ software

Pattern

PN7, PN9, PN15, PN23, User file, Custom

Supported by EasyIQ software

Output Range

1m

Carrier frequency, 50Ω load

0.5

Vrms

500M

Hz

SDG6052X

350M

Hz

SDG6032X

200M

Hz

SDG6022X

WWW.SIGLENT.COM

SDG6000X Pulse/Arbitrary Waveform Generator

PRBS
Parameter
Bit rateSequence length
Rise/fall times

Output Range (Note)

Min.

Max.

Unit

Condition & Note

1u

300M

bps

SDG6052X, SDG6032X

1u

160M

bps

SDG6022X

1n

1u

s

SDG6052X, SDG6032X. 10% ~ 90%, 1 Vpp, 50Ω load

2n

1u

s

SDG6022X. 10% ~ 90%, 1 Vpp, 50Ω load

2m

20

Vpp

≤ 40 Mbps, HiZ load ,

2m

10

Vpp

40 ~ 240 Mbps ( included ), HiZ load

2m

5

Vpp

Above 240 Mbps , HiZ load

Unit

Condition & Note

2

m-1

Typ.

, m = 3, 4, ... , 32

Note : The specification will be divided by 2 while applied to a 50Ω load.

OutputParameter

Min.

Accuracy±(1%+1mVpp)

Amplitude flatness-0.3

Output impedance

49.5

Output current

-200

Typ.

10 kHz sine, 0 V offset+0.3
50

CrosstalkProtection

Max.

dB

50Ω load, 0.5 Vpp, compare to 1MHz Sine
100 kHz sine

50.5

Ω

200

mA

-60

dBc

CH1=CH2=0 dBm, Sine, 50 Ω load

Current limiting, Over voltage protection

Current-limit threshold

±200

mA

±3.5

±4

±4.5

V

The amplitude of the generator <3.2Vpp and the DC offset<|2VDC|

±10.5

±11

±11.5

V

The amplitude of the generator ≥3.2Vpp or the DC offset≥|2VDC|Parameter

Min.

Typ.

Max.

Unit

Condition & Note

CarrierSine, Square, Ramp, Arb

Modulation source

Internal/External

Modulation wave

Sine, Square, Ramp, Noise, Arb

Modulation depth

0

120

%

Modulation frequency

1m

1M

Hz

While modulation source is “Internal”

Max.

Unit

Condition & Note

Over voltage protection
threshold

Modulation
AM

FM
Parameter

Min.

Typ.

CarrierSine, Square, Ramp, Arb

Modulation source

Internal/External

Modulation wave

Sine, Square, Ramp, Noise, Arb

Frequency deviation

00.5*BW

Modulation frequency

1m

1M

Hz

While modulation source is “Internal”

Max.

Unit

Condition & Note

BW is the max. frequency. Limited by frequency settingPM
Parameter

Min.

CarrierSine, Square, Ramp, Arb

Modulation source

Internal/External

Modulation wave

Sine, Square, Ramp, Noise, Arb

Phase deviation

0

360

°

Modulation frequency

1m

1M

Hz

WWW.SIGLENT.COM

Typ.

While modulation source is “Internal”

SDG6000X Pulse/Arbitrary Waveform Generator
ASK
Parameter

Min.

CarrierSine, Square, Ramp, Arb

Typ.

Modulation source

Internal/External

Modulation wave

Square with 50% duty cycle

Keying frequency

1m

Max.

Unit

Condition & Note

1M

Hz

While modulation source is “Internal”

Max.

Unit

Condition & Note

1M

Hz

While modulation source is “Internal”

Max.

Unit

Condition & Note

1M

Hz

While modulation source is “Internal”

Max.

Unit

Condition & Note

1M

Hz

While modulation source is “Internal”

Max.

Unit

Condition & Note

BW is the max. output frequency

FSK
Parameter

Min.

Typ.

CarrierSine, Square, Ramp, Arb

Modulation source

Internal/External

Modulation wave

Square with 50% duty cycle

Keying frequency

1m

PSK
Parameter

Min.

Typ.

CarrierSine, Square, Ramp, Arb

Modulation source

Internal/External

Modulation wave

Square with 50% duty cycle

Keying frequency

1m

PWM
Parameter

Min.

Typ.

Carrier

Pulse

Modulation source

Internal/External

Modulation wave

Sine, Square, Ramp, Noise, Arb

Modulation frequency

1m

Burst
Parameter

Min.

CarrierSine, Square, Ramp, Pulse, Noise, Arb

Typ.

Type

Count (1-1000000 periods), Infinite, Gated

Carrier frequency

2m

BW

Hz

Start/Stop phase

0

360

°

Internal period

1μ

1000

s

Trigger source

Internal, External, Manual

Gated source

Internal/External
100

s

Max.

Unit

Condition & Note

BW is the max. output frequency

Trigger delay

Sweep
Parameter

Min.

Typ.

CarrierSine, Square, Ramp, Arb

Type

Linear, Logarithmic

Direction

Linear: Up, Down, Up & Down
Logarithmic: Up, Down

Carrier frequency

1μ

BW

Hz

Sweep time

1m

500

s

Trigger source

Internal, External, Manual
WWW.SIGLENT.COM

SDG6000X Pulse/Arbitrary Waveform Generator

Frequency Counter
Parameter

Min.

Function

Frequency, Period, Positive/Negative Pulse Width, Duty Cycle

Coupling mode

AC, DC, HF REJ

Frequency range

100m

Input amplitude

Typ.

Max.

Unit

Condition & Note

400M

Hz

DC couplingHz

AC coupling

1

400M

100mVrms±2.5V

DC coupling , < 100 MHz

200mVrms±2.5V

DC coupling , 100 MHz ~ 200MHz

500mVrms±2.5V

DC coupling , Above 200 MHz

100mVrms

5 Vpp

AC coupling , < 100 MHz

200mVrms

5 Vpp

AC coupling , 100 MHz ~ 200MHz

500mVrms

5 Vpp

AC coupling , Above 200 MHz

Input impedance

1MΩReference Clock
10MHz Input
Parameter

Min.

Typ.

Max.

Unit

Condition & Note

Frequency

9.999M

10M

10.001M

Hz

Amplitude

1.4

Vpp

Input impedance5kΩ

AC coupling

Unit

Condition & Note

10M

Hz

Synchronized to internal reference clock

3.3

Vpp

HiZ load50

Ω10MHz Output
Parameter

Min.

Frequency
Amplitude2

Output impedance

Typ.

Max.

Auxiliary In/Out
Trigger Input
Parameter

Min.

Max.

Unit

VIH

2

Typ.

5.5

V

VIL

-0.5

0.8

V

Input impedance

100

kΩ

Pulse width

100

ns

Response time

Condition & Note

1.35

us

Sweep

1.4

us

Burst

Max.

Unit

Condition & NoteV

IOH = –8 mA

V

IOL = 8 mA

Trigger Output
Parameter

Min.

VOH

3.8

Typ.

0.44

VOL
Output impedanceFrequency

WWW.SIGLENT.COM

100

Ω
1

MHz

SDG6000X Pulse/Arbitrary Waveform Generator
Sync OutParameter

Min.

VOH

3.8

Typ.

Max.

0.44

VOL

Unit

Condition & NoteV

IOH = –8 mA

V

IOL = 8 mA

Output impedance100Ω

Pulse width

26.7

ns

Jitter3.3

nsFrequency

10

MHz

Max.

Unit

50

kHz

Peak to peak

Modulation Input
Parameter

Min.

Typ.

Condition & Note

Frequency

0

Input impedance

10

Amplitude @100%
modulation depth

11

12

13

Vpp

Parameter

Min

Typ

Max

Unit

Condition

Voltage

100 - 240 Vrms (± 10%), 50 / 60 Hz
100 - 120 Vrms (± 10%), 400 HzkΩGeneralPower

Power consumption

32.5

50

W

Dual channels, Sine, 1kHz, 10Vpp, 50Ω load

Typ.

Max.

Unit

Condition & Note

Display
Parameter

Min.

Color depth

24

Contrast Ratio

350:1

Luminance

300

Touch Screen Type

bit

cd/m2
Resistive

Environment
Parameter

Min.

Operating temperature
Storage temperature

Max.

Unit

0

40

℃

-20

60

℃

5

90

%

5

50

%

5

95

%

Operating altitude

3048

m

Non –operating altitude

15000

m

Max.

Unit

Operating humidity
Non -operating humidity

Typ.

Condition & Note

≤ 30 ℃
40 ℃≤ 30 ℃

Calibration
Parameter

Min.

Calibration interval

Typ.
1

Condition & Noteyear

MechanicalParameter

Min.

Typ.

Max.

Dimensions

W×H×D = 260.3mm×107.2mm×295.7mm

Unit

Net weight

3.5

kg

Gross weight

4.6

kg

Condition & Note

Compliance
LVD

IEC 61010-1:2010

EMC

EN61326-1:2013
WWW.SIGLENT.COM

SDG6000X Pulse/Arbitrary Waveform Generator

Ordering Information

Product Description
SDG6052X

500 MHz, 2-CH, 2.4 GSa/s, 16-bit

SDG6032X

350 MHz, 2-CH, 2.4 GSa/s, 16-bit

SDG6022X

200 MHz, 2-CH, 2.4 GSa/s, 16-bit

Standard Configurations
Quick start ×1
Power cord ×1
Calibration certificate ×1
USB cable ×1
BNC coaxial cable x2

Optional Configurations
SPA1010

10W Power Amplifier

ATT-20dB

20 dB Attenuator

USB-GPIB

USB-GPIB Adapter

SDG-RMK

Single Instrument Rack Mount Kit

SDG-6000X-IQ

IQ Signal Generator Function

WWW.SIGLENT.COM

SDG6000X
SeriesPulse/Arbitrary
Waveform Generator

About SIGLENTSIGLENT is an international high-tech company, concentrating on R&D, sales,
production and services of electronic test & measurement instruments.

SIGLENT ﬁrst began developing digital oscilloscopes independently in 2002.
After more than a decade of continuous development, SIGLENT has extended
its product line to include digital oscilloscopes, isolated handheld
oscilloscopes, function/arbitrary waveform generators, RF/MW signal
generators, spectrum analyzers, vector network analyzers, digital
multimeters, DC power supplies, electronic loads and other general purpose
test instrumentation. Since its ﬁrst oscilloscope was launched in 2005,
SIGLENT has become the fastest growing manufacturer of digital
oscilloscopes. We ﬁrmly believe that today SIGLENT is the best value inelectronic test & measurement.

Headquarters:
SIGLENT Technologies Co., Ltd
Add: Bldg No.4 & No.5, Antongda Industrial
Zone, 3rd Liuxian Road, Bao'an District,
Shenzhen, 518101, China
Tel: + 86 755 3688 7876
Fax: + 86 755 3359 1582
Email: sales@siglent.com
Website: int.siglent.com
North America:
SIGLENT Technologies America, Inc
6557 Cochran Rd Solon, Ohio 44139
Tel: 440-398-5800
Toll Free: 877-515-5551
Fax: 440-399-1211
Email: info@siglentna.com
Website: www.siglentna.com
Europe:SIGLENT Technologies Germany GmbH
Add: Staetzlinger Str. 70
86165 Augsburg, Germany
Tel: +49(0)-821-666 0 111 0
Fax: +49(0)-821-666 0 111 22
Email: info-eu@siglent.com
Website: www.siglenteu.com

Follow us on
Facebook: SiglentTech