0101010101010101010101
0101010101010101010101
0101010101010101010101
0101010101010101010101
0101010101010101010101
0101010101010101010101
0101010101010101010101
0101010101010101010101

0101010101010101010101
0101010101010101010101
0101010101010101010101
0101010101010101010101
0101010101010101010101
0101010101010101010101
0101010101010101010101
0101010101010101010101

0101010101010101010101
0101010101010101010101
0101010101010101010101
0101010101010101010101 Digital
Oscilloscope
0101010101010101010101
0101010101010101010101 Waveform
Generator
0101010101010101010101 RF0101010101010101010101
Signal Generator
0101010101010101010101
0101010101010101010101 Spectrum
Analyzer
Every Bench.
0101010101010101010101
Vector0101010101010101010101
Network Analyzer
0101010101010101010101
0101010101010101010101
Every Engineer.
DC Power Supply
0101010101010101010101
0101010101010101010101 DC
Electronic Load
Every Day.

Product Selection Guide

Digital Multimeter
Probes & Accessories

SIGLENT TECHNOLOGIES CO., LTD

Super Phosphor Oscilloscope

SDS7000A Series

SDS6000A Series

SDS7404A H12 (4 GHz)
SDS7304A H12 (3 GHz)

SDS6204A (2 GHz)
SDS6104A (1 GHz)
SDS6054A (500 MHz)

Features and Benefits:• 4 analog channels, up to 4 GHz bandwidth with up to 20 GSa/s sample rate
• 12-bit ADC• Low background noise: 220 μVrms @ 4 GHz bandwidth
SPO technology• Waveform capture rates up to 1,000,000 wfm/s• Supports 256-level intensity grading and color temperature display modes• 500 Mpts/ch standard, 1 Gpts/ch optional• Digital trigger system• Intelligent trigger: Edge, Slope, Pulse, Window, Runt, Interval, Dropout,
Pattern, Qualified, Nth edge, Setup/hold, Delay and Video (HDTV supported).
Zone Trigger simplifies advanced triggering• Serial bus triggering and decoder, supports protocols I2C, SPI, UART, CAN,
LIN, CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT, Manchester, ARINC429 andUSB 2.0• Segmented acquisition (Sequence) mode, dividing the maximum record
length into multiple segments (up to 124,000), according to trigger conditions
set by the user, with a very small dead time between segments to capture thequalifying event• History waveform record (History) function, the maximum recorded waveform
length is 124,000 frames• Automatic measurements on 50+ parameters, supports statistics withhistogram, track, trend, Gating measurement, and measurements on Math,History and Memory traces• 4 Math traces (32 Mpts FFT, Filter, addition, subtraction, multiplication,
division, integration, differential, square root, etc.), supports formula editor• Abundant data analysis functions such as Search, Navigate, SignalScan,
Digital Voltmeter, Counter, Waveform Histogram, Bode plot, Power Analysis,
Eye/Jitter Analysis and Compliance Test
• High Speed hardware-based Average, Hi-Res; High Speed hardware-based
Mask Test function, with Mask Editor tool for creating user-defined masks• 16 digital channels (optional)• Built-in 50 MHz waveform generator• Large 15.6” HD TFT-LCD display with 1920 * 1080 resolution; Capacitive
touch screen supports multi-touch gestures• Interfaces include: 4x USB Host 3.1 Gen 1, 2x USB 3.0 Host, USB 2.0 Device,
2x 1000M LAN, DVI-D, DP 1.2, HDMI 1.4, Audio, External Triger In, Aux Out(Pass/Fail, Trigger Out), 10 MHz In, 10 MHz Out• Built-in web server supports remote control over the LAN port using a web
browser. Supports SCPI remote control commands. Supports external mouseand keyboard

Features and Benefits:• 4 analog channels, up to 2 GHz bandwidth with 5 GSa/s (10 GSa/s ESR)
sample rate at each channel• Low background noise, supports 0.5 mV/div to 10 V/div vertical scales• SPO technology• Waveform capture rates up to 170,000 wfm/s (normal mode), and 750,000
wfm/s (sequence mode)• Supports 256-level intensity grading and color temperature display modes• 500 Mpts Record length in total for all 4 channels• Digital trigger system• Intelligent trigger: Edge, Slope, Pulse, Window, Runt, Interval, Dropout,
Pattern, Qualified, Nth edge, Setup/hold, Delay and Video (HDTV supported).
Zone Trigger simplifies advanced triggering• Serial bus triggering and decoder, supports protocols I2C, SPI, UART, CAN,
LIN, CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT and Manchester• Segmented acquisition (Sequence) mode, dividing the maximum record
length into multiple segments (up to 80,000), according to trigger conditions
set by the user, with a very small dead time between segments to capture thequalifying event• History waveform record (History) function, the maximum recorded waveform
length is 80,000 frames• Automatic measurements on 50+ parameters, supports statistics withhistogram, track, trend, Gating measurement, and measurements on Math,History and Ref• 4 Math traces (8 Mpts FFT, addition, subtraction, multiplication, division,
integration, differential, square root, etc.), supports formula editor• Abundant data analysis functions such as Search, Navigate, Digital Voltmeter,
Counter, Waveform Histogram, Bode plot, Power Analysis and Eye/Jitter
Analysis• High Speed hardware-based Average, Hi-Res; High Speed hardware-based
Mask Test function, with Mask Editor tool for creating user-defined masks• 16 digital channels (optional)• 25 MHz function / arbitrary waveform generator, built-in multiple predefined
waveforms• Large 12.1” TFT-LCD display with 1280 * 800 resolution; Capacitive touch
screen supports multi-touch gestures• Interfaces include: USB Hosts, USB Device (USBTMC), LAN (VXI-11/Telnet/
Socket), micro SD card, Pass/Fail, Trigger Out, HDMI
• Built-in web server supports remote control over the LAN port using a web
browser. Supports SCPI remote control commands. Supports external mouseand keyboard

www.siglent.com 1

Super Phosphor Oscilloscope

SDS6000L Series
SDS6208L (2 GHz)
SDS6108L (1 GHz)
SDS6058L (500 MHz)

SDS5000X Series
SDS6204L (2 GHz)
SDS6104L (1 GHz)
SDS6054L (500 MHz)SDS5104X (1 GHz)
SDS5054X (500 MHz)
SDS5034X (350 MHz)

Features and Benefits:

Features and Benefits:• 8/4 analog channels + 1 external trigger. Designed for expansion. Combine
multiple units for a high-speed acquisition system with up to 512 channels.

• 1 GHz, 500 MHz, 350 MHz models with real-time sample rate up to 5 GSa/s• SPO technology• Up to 2 GHz bandwidth with 5 GSa/s (10 GSa/s ESR) sample rate at eachchannel• Waveform capture rates up to 110,000 wfm/s (normal mode), and 500,000• Low background noise, supports 0.5 mV/div to 10 V/div vertical scales• SPO technology• Supports 256-level intensity grading and color temperature display modes• Waveform capture rates up to 170,000 wfm/s (normal mode), and
750,000 wfm/s (sequence mode)• Supports 256-level intensity grading and color temperature display modes• 500 Mpts Record length in total for all 4 channels• Digital trigger system• Intelligent trigger: Edge, Slope, Pulse, Window, Runt, Interval, Dropout,
Pattern, Qualified, Nth edge, Setup / hold, Delay and Video (HDTV
supported). Zone Trigger simplifies advanced triggering• Serial bus triggering and decoder, supports protocols I2C, SPI, UART, CAN,
LIN, CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT and Manchester• Segmented acquisition (Sequence) mode, dividing the maximum record
length into multiple segments (up to 80,000), according to trigger conditions
set by the user, with a very small dead time between segments to capture thequalifying event

wfm/s (sequence mode)• Record length up to 250 Mpts/ch, 500 Mpts in total for all 4 channels• Digital trigger system• Intelligent trigger: Edge, Slope, Pulse, Window, Runt, Interval, Dropout,
Pattern, Qualified, Nth edge, Setup/hold, Delay and Video (HDTV supported).
Trigger zone helps to simplify advanced triggering• Serial bus triggering and decoder, supports protocols I2C, SPI, UART, CAN,
LIN, CAN FD, FlexRay, I2S and MIL-STD-1553B, SENT and Manchester• Low background noise, supports 0.5 mV/div to 10 V/div voltage scales• Segmented acquisition (Sequence) mode, dividing the maximum record
length into multiple segments (up to 100,000), according to trigger conditions
set by the user, with a very small dead time between segments to capture thequalifying event• History waveform record (History) function, the maximum recorded waveform
length is 100,000 frames• Automatic measurement function on 50+ parameters, supports statistics• History waveform record (History) function, the maximum recorded waveform
length is 80,000 frameswith histogram,trend, Gating measurement, Math measurement, History• Automatic measurements on 50+ parameters, supports statistics withhistogram, track, trend, Gating measurement, and measurements on Math,• Math function (2 Mpts FFT, addition, subtraction, multiplication, division,

History, Memory and Ref• 4 Math traces (8 Mpts FFT, addition, subtraction, multiplication, division,
integration, differential, square root, etc.), supports formula editor

measurement and Ref measurement
integration, differential, square root, etc.), supports formula editor
Abundant data analysis functions such as Search, Navigate, Digital Voltmeter,
Counter, Waveform Histogram, Bode plot and Power Analysis• High Speed hardware-based Average, ERES (Enhanced Resolution)• Abundant data analysis functions such as Search, Navigate, Digital Voltmeter,
Counter, Waveform Histogram, Power Analysis and Eye/Jitter Analysis• High Speed hardware-based Mask Test function, with Mask Editor tool for• 16 digital channels (optional)• 25 MHz function / arbitrary waveform generator, built-in multiple predefined• 16 digital channels (optional) with sample rate up to 1.25 GSa/s, record

waveforms• Interfaces include: 4x USB Hosts, USB Device (USBTMC), 1000M LAN (VXI11/ Telnet/ Socket), micro SD card, Pass/Fail, Trigger Out, HDMI, 10MHz In,
10MHz Out• Built-in web server supports remote control over the LAN port using a web
browser. Supports SCPI remote control commands. Supports external mouseand keyboard

creating user-defined masks
length up to 62.5 Mpts
• 25 MHz function / arbitrary waveform generator, built-in multiple predefined
waveforms• Large 10.1” TFT-LCD display with 1024 * 600 resolution; Capacitive touch
screen supports multi-touch gestures• Supports external mouse and keyboard• 10 types of one-button shortcuts• Multiple interfaces: USB Host, USB Device (USBTMC), LAN (VXI-11, telnet,
socket, web), Pass/Fail, Trigger Out, 10 MHz In, 10 MHz Out, VGA output• Built-in web server supports remote control by the LAN port using a web
browser• Supports SCPI remote control commands

www.siglent.com 2

Super Phosphor Oscilloscope

SDS3000X HD Series

SDS2000X HD Series

SDS3104X HD (1 GHz)
SDS3054X HD (500 MHz)SDS3034X HD (350 MHz)SDS2354X HD (350 MHz)SDS2204X HD (200 MHz)SDS2104X HD (100 MHz)

Features and Benefits:

Features and Benefits:• 4 analog channels, up to 1 GHz bandwidth with up to 4GSa/s sample rate

• 12-bit High Resolution• 12-bit ADC• 12-bit Analog-Digital Convertors with sample rate up to 2 GSa/s• Low background noise: 125 μVrms @ 1 GHz bandwidth• Front ends with 70 μVrms noise floor @ 500 MHz bandwidth and 0.5% DC• SPO technology• Waveform capture rates up to 200,000 wfm/s in Normal mode and 890,000
wfm/s in Sequence mode
• Supports 256-level intensity grading and color temperature display modes• Up to 400 Mpts/ch record length• Digital trigger system• Intelligent trigger: Edge, Slope, Pulse, Window, Runt, Interval, Dropout,
Pattern, Qualified, Nth edge, Setup/hold, Delay and Video (HDTV supported).
Zone Trigger simplifies advanced triggering• Serial bus triggering and decoder, supports protocols I2C, SPI, UART, CAN,
LIN, CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT, Manchester and ARINC429• Segmented acquisition (Sequence) mode, dividing the maximum record

gain accuracy• 4 analog channels, up to 350 MHz bandwidth (upgradable to 500 MHz)• SPO technology• Waveform capture rate up to 100,000 wfm/s (normal mode), and500,000 wfm/s (sequence mode)• Supports 256-level intensity grading and color temperature display modes• Up to 200 Mpts/ch record length• Digital trigger system• Intelligent trigger: Edge, Slope, Pulse, Window, Runt, Interval, Dropout,
Pattern, Qualified, Nth edge, Setup/hold, Delay and Video (HDTV supported).
Zone Trigger simplifies advanced triggering• Serial bus triggering and decoder, supports protocols I2C, SPI, UART, CAN,

length into multiple segments (up to 80,000), according to trigger conditions

LIN, CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT and Manchester

set by the user, with a very small dead time between segments to capture the• Segmented acquisition (Sequence) mode, dividing the maximum record

qualifying event• History waveform record (History) function, the maximum recorded waveform
length is 80,000 frames• Automatic measurements on 50+ parameters, supports statistics withhistogram, track, trend, Gating measurement, and measurements on Math,History and Memory traces• 4 Math traces (4 Mpts FFT, Filter, addition, subtraction, multiplication, division,
integration, differential, square root, etc.), supports formula editor• Abundant data analysis functions such as Search, Navigate, Digital Voltmeter,
Counter, Waveform Histogram, Bode plot and Power Analysis• High Speed hardware-based Average, ERES; High Speed hardware-based
Mask Test function, with Mask Editor tool for creating user-defined masks• 16 digital channels (optional)• External 50 MHz waveform generator (optional)• Large 10.1” TFT-LCD display with 1024 * 600 resolution; Capacitive touch
screen supports multi-touch gestures• Interfaces include: 2x USB 3.0 Host, 1x USB 2.0 Host , USB 3.0 Device(USBTMC), 1000M LAN ((VXI-11/Telnet/Socket/LXI)), External Triger In, Aux
Out (Pass/Fail, Trigger Out)• Built-in web server supports remote control over the LAN port using a web
browser. Supports SCPI remote control commands. Supports external mouseand keyboard

length into multiple segments (up to 80,000), according to trigger conditions
set by the user, with a very small dead time between segments to capture thequalifying event• History waveform record (History) function, the maximum recorded waveform
length is 80,000 frames• Automatic measurements on 50+ parameters, supports statistics withhistogram, track, trend, Gating measurement, and measurements on Math,History and Ref• 2 Math traces (2 Mpts FFT, addition, subtraction, multiplication, division,
integration, differential, square root, etc.), supports formula editor• Abundant data analysis functions such as Search, Navigate, Digital Voltmeter,
Counter, Waveform Histogram, Bode plot and Power Analysis• High Speed hardware-based Average, ERES; High Speed hardware-based
Mask Test function, with Mask Editor tool for creating user-defined masks• 16 digital channels (optional)• Built-in 25 MHz waveform generator• Large 10.1” TFT-LCD display with 1024 * 600 resolution; Capacitive touch
screen supports multi-touch gestures• Interfaces include: USB Hosts, USB Device (USBTMC), LAN (VXI-11/Telnet/
Socket), Pass/Fail, Trigger Out• Built-in web server supports remote control over the LAN port using a web
browser. Supports SCPI remote control commands. Supports external mouseand keyboard

www.siglent.com 3

Super Phosphor Oscilloscope

SDS2000X Plus Series

SDS2000X-E Series

SDS2354X Plus (350 MHz)SDS2204X Plus (200 MHz)SDS2104X Plus / SDS2102X Plus (100 MHz)SDS2352X-E (350 MHz)SDS2202X-E (200 MHz)

Features and Benefits:• 350 MHz, 200 MHz, 100 MHz models with real-time sample rate up to 2 GSa/s.
A 500 MHz bandwidth upgrade option is available for 350 MHz models.
• SPO technology• Waveform capture rates up to 120,000 wfm/s (normal mode) and 500,000
wfm/s (sequence mode)• Supports 256-level intensity grading and color temperature display modes• Record length up to 200 Mpts/ch, 400 Mpts in total for all 4 channels• Digital trigger system• Intelligent trigger: Edge, Slope, Pulse, Window, Runt, Interval, Dropout,
Pattern and Video (HDTV supported). Trigger zone helps to simplify advanced
triggering

Features and Benefits:• Real-time sampling rate up to 2 GSa/s (1 GSa/s per channel, if both channels
active)• Intelligent trigger: Edge, Slope, Pulse Width, Window, Runt, Interval, Timeout (Dropout), Pattern• Serial bus triggering and decoding (standard), supports protocols IIC, SPI,UART, CAN, LIN• Video trigger, supports HDTV
• Low background noise with voltage scales from 500 μV/div to 10 V/div• 10 types of one-button shortcuts, supports Auto Setup, Default, Cursors,
Measure, Roll, History, Display/Persist, Clear Sweep, Zoom and Print• Segmented acquisition (Sequence) mode, divides the maximum record length
into multiple segments (up to 80,000), according to trigger conditions set by• Serial bus triggering and decoder, supports I2C, SPI, UART, CAN, LIN

the user, with a very small dead time segment to capture qualifying events(Standard) and CAN FD, FlexRay, I2S, and MIL-STD-1553B, SENT and• History waveform record (history) function (maximum recorded waveform
length is 80,000 frames)

Manchester (optional) protocols• Low background noise, features 0.5 mV/div to 10 V/div voltage scales• 10-bit mode provides higher resolution and lower noise• Segmented acquisition (Sequence) mode, dividing the maximum record
length into multiple segments (up to 90,000), according to trigger conditions
set by the user, with a very small dead time between segments to capture thequalifying event• History waveform record (History) function for up to 90,000 triggered
waveforms (frames)• Automatic measurement function on 50+ parameters, supports statistics withhistogram and trend• Two Math traces, support 2 Mpts FFT, +, -, x, ÷, d/dt, ∫dt, √, average, ERES,
and formula editor• Abundant data processing and analysis functions such as Search,
Navigate, Mask Test, Bode plot, Power Analysis (optional) and Counter• 16 digital channels (optional)• Built-in 50 MHz waveform generator (optional)• Large 10.1” TFT-LCD display with 1024x600 resolution; Capacitive touch
screen supports multi-touch gestures• Multiple interfaces: USB Host, USB Device (USBTMC), LAN(VXI-11/Telnet/
Socket), Pass/Fail, Trigger Out• Built-in web server supports remote control by the LAN port using a web
browser; Supports SCPI remote control commands

www.siglent.com 4• Automatic measurement function for 38 parameters as well as Measurement
Statistics, Zoom, Gating, Math, History and Reference functions• 1 Mpts FFT• Math and measurement functions use all sampled data points in memory (upto 28 Mpts)• Preset key can be customized for user settings or factory “defaults”• Security Erase mode
• High Speed hardware based Pass / Fail function• Search and navigate• Large 7 inch TFT -LCD display with 800 * 480 resolution• Multiple interface types: USB Host, USB Device (USB -TMC), LAN, Pass / Fail,
Trigger Out• Supports SCPI remote control commands• VXI-11+SCPI, Telnet (port 5024) +SCPI and Socket (port 5025) +SCPI
programming over LAN• Supports web control and virtual instrument control panel for both PC andmobile terminals• Web control update rate of up-to 10 times/s provides nearly real-time update
rate• Supports Multi-language display and embedded online help

Super Phosphor Oscilloscope

SDS1000X HD Series

SDS800X HD Series

SDS1204X HD / SDS1202X HD (200 MHz)SDS1104X HD / SDS1102X HD (100 MHz)SDS824X HD / SDS822X HD (200 MHz)SDS814X HD / SDS812X HD (100 MHz)
SDS804X HD / SDS802X HD (70 MHz)

Features and Benefits:• 12-bit High Resolution• 12-bit Analog-Digital Convertors with sample rate up to 2 GSa/s• Front ends with 70 μVrms noise floor @ 200 MHz bandwidth• 2/4 analog channels, up to 200 MHz bandwidth• SPO technology• Waveform capture rate up to 120,000 wfm/s (normal mode), and 500,000
wfm/s (sequence mode)• Supports 256-level intensity grading and color temperature display modes

Features and Benefits:• 12-bit High Resolution• 12-bit Analog-Digital Convertors with sample rate up to 2 GSa/s• Front ends with 70 μVrms noise floor @ 200 MHz bandwidth• 2/4 analog channels, up to 200 MHz bandwidth• SPO technology• Waveform capture rate up to 120,000 wfm/s (normal mode), and 500,000
wfm/s (sequence mode)• Up to 100 Mpts record length• Supports 256-level intensity grading and color temperature display modes• Digital trigger system• Up to 100 Mpts record length• Intelligent trigger: Edge, Slope, Pulse width, Window, Runt, Interval, Dropout,
Pattern, Video (HDTV supported) , Qualified, Nth edge, Delay, Setup/Holdtime.
• Serial bus triggering and decoder, supports protocols I2C, SPI, UART, CAN,
LIN, CAN FD(decode only), FlexRay(decode only)• Segmented acquisition (Sequence) mode, dividing the maximum record
length into multiple segments (up to 80,000), according to trigger conditions• Digital trigger system• Intelligent trigger: Edge, Slope, Pulse width, Window, Runt, Interval, Dropout,
Pattern, Video (HDTV supported) , Qualified, Nth edge, Delay, Setup/Holdtime.
• Serial bus triggering and decoder, supports protocols I2C, SPI, UART, CAN,
LIN
• Segmented acquisition (Sequence) mode, dividing the maximum record

set by the user, with a very small dead time between segments to capture thelength into multiple segments (up to 80,000), according to trigger conditions

qualifying event

set by the user, with a very small dead time between segments to capture the• History waveform record (History) function, the maximum recorded waveform
length is 80,000 frames• Automatic measurements on 50+ parameters, supports statistics withhistogram, track, trend, Gating measurement, and measurements on Math,History and Ref• 4 Math traces (2 Mpts FFT, addition, subtraction, multiplication, division,
integration, differential, square root, etc.), supports formula editor• Abundant data analysis functions such as Search, Navigate, Counter, Bodeplot and Power Analysis• High Speed hardware-based Mask Test function, with Mask Editor tool for
creating user-defined masks• 16 digital channels (optional)

qualifying event• History waveform record (History) function, the maximum recorded waveform
length is 80,000 frames• Automatic measurements on 50+ parameters, supports statistics withhistogram, track, trend, Gating measurement, and measurements on Math,History and Ref• 4 Math traces (2 Mpts FFT, addition, subtraction, multiplication, division,
integration, differential, square root, etc.), supports formula editor• Abundant data analysis functions such as Search, Navigate, Counter, Bodeplot and Power Analysis• High Speed hardware-based Mask Test function, with Mask Editor tool for
creating user-defined masks• 25 MHz waveform generator(optional)• 16 digital channels (optional)• Large 10.1” TFT-LCD display with 1024 * 600 resolution; Capacitive touch• 25 MHz waveform generator(optional)

screen supports multi-touch gestures• Interfaces include: USB Hosts, USB Device (USBTMC), LAN (VXI-11/Telnet/
Socket), Pass/Fail, Trigger Out• Built-in web server supports remote control over the LAN port using a web
browser. Supports SCPI remote control commands. Supports external mouseand keyboard• 7” TFT-LCD display with 1024 * 600 resolution; Capacitive touch screen
supports multi-touch gestures• Interfaces include: USB Hosts, USB Device (USBTMC), LAN (VXI-11/Telnet/
Socket), Pass/Fail, Trigger Out• Built-in web server supports remote control over the LAN port using a web
browser. Supports SCPI remote control commands. Supports external mouseand keyboard. Supports NTP.

www.siglent.com 5

Digital Storage Oscilloscope

SDS1000X-E Series
SDS1204X-E / SDS1202X-E (200 MHz)SDS1104X-E (100 MHz)
Features and Benefits:• Two channel series have one 1 GSa/s ADC, four channel series have two 1
GSa/s ADCs. When all channels are enabled, each channel has a maximum
sample rate of 500 MSa/s. When a single channel per ADC is active, it has
sampling rate of 1 GSa/s• The newest generation of SPO technology• Waveform capture rate up to 100,000 wfm/s (normal mode), and400,000 wfm/s (sequence mode)• Supports 256-level intensity grading and color display modes• Record length up to 14 Mpts
• Digital trigger system• Intelligent trigger: Edge, Slope, Pulse Width, Window, Runt, Interval, Timeout (Dropout), Pattern• Serial bus triggering and decoding (Standard), supports protocols IIC, SPI,UART, RS232, CAN, LIN• Segmented acquisition (Sequence) mode, divides the maximum record length
into multiple segments (up to 80,000), according to trigger conditions set by
the user, with a very small dead time segment to capture the qualifying event.
• 1 Mpts FFT• Math and measurement functions use all sampled data points (up to 14 Mpts)• MSO, 16 digital channels (four channel series only, optional)• Search and navigate (four channel series only)• USB AWG module (four channel series only, optional)• Bode plot (four channel series only)

www.siglent.com 6

SDS1000DL+ / SDS1000CML+ Series
SDS1152CML+ (150 MHz)SDS1102CML+ (100 MHz)SDS1072CML+ (70 MHz)SDS1052DL+ (50 MHz)
Features and Benefits• 50 MHz, 70 MHz, 100 MHz, 150 MHz bandwidth models• Real-time sampling rate up to 1 GSa/s, Equivalent-time sampling rate up to
50 GSa/s• Memory Depth up to 2 Mpts
• Trigger types: Edge, Pulse, Video, Slope, Alternate• Waveform math functions:+, -, *, /, FFT• 6 digits frequency counter• Supports Multi-language display and embedded online help
• Screensaver from 1 minute to 5 hours• Digital filter and waveform recorder function• 7 inch TFT-LCD display with 800 * 480 resolution

Handheld Oscilloscope

Arbitrary Waveform Generator

SDG7000A Series
SHS1000X/SHS800X Series
SHS1202X (200 MHz)SHS1102X (100 MHz)SHS820X (200 MHz)SHS810X (100 MHz)
Features and Benefits• 200 MHz, 100 MHz bandwidth models• Sample rate of 1 GSa/s (single-channel), Sample rate of 500 MSa/s (twochannels)• The Siglent SPO technology• Waveform capture rates up to 100,000 wfm/s (normal mode) and
400,000 wfm/s (sequence mode)• Supports 256-level intensity grading and color temperature display modes• Record length up to 12 Mpts
• Digital trigger system• Intelligent trigger: Edge, Slope, Pulse Width, Window, Runt, Interval, Timeout (Dropout), Pattern• Serial bus triggering and decoding (Standard) for IIC, SPI, UART, CAN, and
LIN protocols• Video trigger/HDTV• Low background noise with voltage scales from 2 mV/div to 100 V/div• 3 one-button shortcuts for Oscilloscope, Multimeter and Recorder functions• 8 one-button shortcuts for: Run/Stop, Auto Setup, Default, Measure, Cursors,
Display/Persist, Clear Sweep and Print. More function shortcuts available
when combined with the shift button• Segmented acquisition (Sequence) mode, divides the maximum record length
into multiple segments (up to 80,000), according to trigger conditions set by
the user, with a very small dead time segment to capture the qualifying event• History waveform record (History) function (maximum recorded waveform
length is 80,000 frames)

SDG7102A (1 GHz)
SDG7052A (500 MHz)
SDG7032A (350 MHz)
Features and Benefits• Dual channel differential/single-ended output, 16-bit LVDS/LVTTL digital bus
output• High-performance sampling system with 5 GSa/s sample rate and 14 -bit
vertical resolution• 1 GHz maximum bandwidth• Generates arbitrary waveform with sample rates of 0.01 Sa/s ~ 2.5 GSa/s,
with maximum memory depth of 512 Mpts, and provides segment editing /playback functions• Generates vector signals with up to 500 MS/s symbol rate
• Generates low jitter pulses with 1 ns minimum pulse width and 500 ps
minimum edge
• Up to 1 GHz bandwidth White Gaussian Noise and the bandwidth is adjustable• Supports PRBS up to 312.5 Mbps
• The digital bus can output digital signals up to 1 Gbps.
• Supports analog/digital modulation, sweeping and bursting• Enhanced dual channel operation functions: inter channel tracking, coupling
and copying; Dual channel superposition function; Supports mutual
modulation between channels• The 24 Vpp analog output is superimposed with ± 12 Vdc offset to provide amaximum output range of ± 24 V (48 V)• High precision Frequency Counter• 5-inch capacitive touch screen with resolution of 800x480; Supports
external mouse and keyboard operation; Supports WebServer to control the
instruments remotely• Supports multiple interfaces: 10 MHz In, 10 MHz Out, Trigger In/Out, Markers
etc• Supports SCPI command for easy integration into test systems• Automatic measurement function for 38 parameters as well as Measurement
Statistics, Zoom, Gating, Math, History and Reference functions• 1 Mpts FFT. Support Peaks and Markers• Math and measurement functions use all sampled data points (up to 12 Mpts)• Math functions (FFT, addition, subtraction, multiplication, division, integration,differential, square root)• Default key can be customized for user settings or factory “defaults”• Supports Multi-language display and embedded online help
• Security Erase mode
• Search and navigate function• Includes Recorder mode, including Sample and Measurement Loggers• 6000 counts Digital Multimeter, Support DCV, ACV, DCI, ACI, Resistance,
Diode, Capacitance, Continuity test.
• True RMS AC Voltage/Current measurement multimeter• 5.6-inch TFT-LCD display with 640 * 480 resolution• Interface types: Isolated USB Host, USB Device (MicroUSB -TMC)• Supports SCPI remote control commands• UL2054 certified lithium battery pack, 6900 mAh capacity, external charger• IP Rating: IP51
• Compliance with UL61010-1, UL61010-2-030, UL61010-2-033
www.siglent.com 7

Function/Arbitrary Waveform Generator

SDG6000X Series

SDG2000X Series

SDG6052X (500 MHz)
SDG6032X (350 MHz)
SDG6022X (200 MHz)

SDG2122X (120 MHz)
SDG2082X (80 MHz)
SDG2042X (40 MHz)

Features and Benefits

Features and Benefits• Innovative TrueArb and EasyPulse technology• Dual-channel, 120 MHz maximum bandwidth, 20 Vpp maximum output• Dual-Channel, 500 MHz maximum bandwidth, 20 Vpp maximum output

amplitude, high fidelity output with 80 dB dynamic range• High-performance sampling system with 1.2 GSa/s sampling rate and 16-bit

amplitude, high fidelity output with 80 dB dynamic range• High-performance sampling system with 2.4 GSa/s sampling rate and 16-bit
vertical resolution• Multi-function signal generator, meeting requirements in wide range,
Continuous Wave Generator, Pulse Generator, Function Arbitrary Waveform
Generator, IQ Signal Generator (optional), Noise Generator, PRBS Generator• Sweep and Burst function• Harmonics function• Waveform Combining function• Channel Coupling, Copy and Tracking function• 196 built-in arbitrary waveforms• High precision Frequency Counter• Standard interfaces include: USB Host, USB Device (USBTMC), LAN (VXI-11,
Socket, Telnet), GPIB (Optional)• 4.3” touch screen display for easier operationwww.siglent.com 8

vertical resolution. No detail in your waveforms will be lost
• Innovative TrueArb technology, based on a point-by-point architecture,
supports any 8 pts~8 Mpts Arb waveform with a sampling rate in range of
1 μSa/s~75 MSa/s• Innovative EasyPulse technology, capable of generating lower jitter Square or
Pulse waveforms, brings a wide range and extremely high precision in pulse
width and rise/fall time adjustments• Sweep and Burst function; Harmonics mode supported• 4.3” touch screen display for easier operation

Function/Arbitrary Waveform Generator

SDG1000X Series
SDG1062X (60 MHz)
SDG1032X (30 MHz)
Features and Benefits• 150 MSa/s sampling rate, 14-bit vertical resolution, and 16 kpts waveform
length• Innovative TrueArb and EasyPulse technology• Special circuit for Square wave function, can generate Square waves up to 60
MHz with jitter less than 300 ps+0.05 ppm of period• Plenty of analog and digital modulation types: AM, DSB-AM, FM, PM, FSK,
ASK, PSK and PWM, Sweep and Burst functions• Waveform Combining function• High precision Frequency Counter• Standard interfaces: USB Host, USB Device (USBTMC), LAN (VXI-11)• Optional interface: GPIB

Programmable Switching DC Power Supply

SPS5000X Series
SPS5161X (160 V, 7.5 A)
SPS5162X (160 V, 15 A)
SPS5163X (160 V, 22.5 A)
SPS5164X (160 V, 7.5 A)
SPS5165X (160 V, 7.5 A)
SPS5051X (50 V, 10 A)
SPS5041X (40 V, 30 A)
SPS5042X (40 V, 60 A)
SPS5043X (40 V, 90 A)
SPS5044X (40 V, 30 A)
SPS5045X (40 V, 30 A)

SPS5081X (80 V, 15 A)
SPS5082X (80 V, 30 A)
SPS5083X (80 V, 45 A)
SPS5084X (80 V, 15 A)
SPS5085X (80 V, 15 A)

Features and Benefits• Rated Output Voltage: 40 V, 50 V, 80 V, 160 V• Rated Output Power: 180 W, 360 W, 720 W, 1080 W• Wide range of output voltage and current, high efficiency power supply• CV, CC priority mode selection, better protection of equipment under test
• Load transient recovery time (Load change from 50~100%) <1 ms• Adjustable slew rate of output voltage and current• Setting and readback resolution: 1 mV, 1 mA
• User enabled internal output discharge circuit to accelerate the down
programming of the output voltage• Remote Voltage Sensing• List function up to 50 steps; can be created from the front panel or byimporting list sequence files from a USB memory device• External analog voltage and resistor control of voltage or current output• External voltage and current monitoring output• OVP, OCP, LPP, OTP protection.
• 2.4-inch OLED high brightness liquid crystal display, 170-degree viewing angle• Standard Interface: USB, LAN, Analog Control Interface

SDG800 Series
SDG830 (30 MHz)
SDG810 (10 MHz)

• Optional Interface: USB-GPIB module• 1/2, 1/3, 1/6 rack mount size
• Embedded Web Server offers remote control through a web browser without
the need for the driver or software

Features and Benefits• Advanced DDS technology, 3.5 inch color TFT-LCD• 125 MSa/s sampling rate,14 bit vertical resolution,16 Kpts max wave length• 5 types of standard waveforms, built-in 46 types of arbitrary waveforms, sync
signal output, 1 μHz frequency resolution• Complete modulation functions: AM,DSB-AM, FM, PM, FSK, ASK, PWM, linear/
logarithmic sweep and burst• Innovative EasyPulse technology, can output pulse of low jitter, quick rising/falling edge
• Support USB-TMC protocol and SCPI programming command control• Arbitrary waveform edit software, provides lots of painting method, capableof edit complicate waveform quickly and precisely
www.siglent.com 9

Programmable Linear DC Power Supply

Programmable Linear DC Power Supply

SPD3303X Series
SPD3303X-E (10 mv, 10 mA)
SPD3303X (1 mv, 1 mA)
Features and Benefits• 3 independent controlled and isolated outputs, 32 V/3.2 A×2,
2.5 V/3.3 V/5 V/3.2 A×1, total 220 W• 5 digits Voltage, 4 digits Current Display, Minimum Resolution: 1 mV/1 mA(SPD3303X)• Supports front panel timing output functions• 4.3 inch true color TFT- LCD 480x272 display• 3 types of output modes: independent, series, parallel• 100 V/120 V/220 V/230 V compatible design to meet the needs of different
power grids• Intelligent temperature-controlled fan, effectively reducing noise• Clear graphical interface, with the waveform display function• Internal 5 groups of system parameter save/recall, supports data storage
space expansion• Provides PC software: Easypower, supports SCPI, LabView driver

SPD1000X Series
SPD1168X (16 V/8 A)
SPD1305X (30 V/5 A)
Features and Benefits• Single path high-precision programmable voltage output
16 V/8 A, total power up to 128 W
30 V/5 A, total power up to 150 W• Stable, reliable, Low ripple and noise：≤ 350 uVrms/3 mVpp; < 2 mArms• Fast transient response time: < 50 μs• 5 digit Voltage, 4 digit Current Display, Minimum Resolution: 1 mV/1 mA• Supports front panel timing output functions• 2.8 inch true color TFT- LCD 240 *320 display• 2 types of output modes：Two-wire output mode, 4-wire compensation output
mode, Maximum compensation voltage 1 V• 100/120/220/230 V compatible design to meet the needs of different power
grids• Intelligent temperature-controlled fan reduces noise• Clear graphical interface, with the waveform display function• Internal 5 groups of system parameter save/recall• Includes PC software: Easypower, supports SCPI, LabView driver

SPD3303C Series
SPD3303C (10 mv,10 mA)
Features and Benefits• 3 independent controlled and isolated output: 32 V/3.2 A×2, 2.5 V/3.3 V/
5 V/3.2 A×1, total power 220 W• Resolution: 10 mV/10 mA• Supports panel timing output functions• LED display; 32 V full scale, 4 digits; 3.2 A full scale, 3 digits• 100 V/120 V/220 V/230 V compatible design, to meet the needs of different
power grids• Intelligent fan for temperature control, effectively reduces noise• Five groups of internal system parameter storage. Supports data storage
space expansion• Supports SCPI commands & USB device interface. Includes PC software

www.siglent.com 10

Programmable DC Electronic Load

SDL1000X Series
SDL1030X/X-E (300 W)
SDL1020X/X-E (200 W)
Features and Benefits

Digital Multimeter

SDM3065X Series
SDM3065X
SDM3065X-SC (with Scanner Card)
Features and Benefits• SDL1020X (Single channel ): DC 150 V/30 A, total power up to 200 W•

4.3” TFT-LCD, 480*272• SDL1030X (Single channel ): DC 150 V/30 A, total power up to 300 W•

Real 6½ digits readings resolution (2,200,000 counts)• 4 static modes / Dynamic mode: CC/CV/CR/CP•

1Gb Nand flash size, Mass storage configuration files and data files• CC Dynamic mode: Continuous, pulsed, toggled•

True-RMS AC Voltage and AC Current measuring• CC Dynamic mode: 25 kHz, CP Dynamic mode: 12.5 kHz, CV Dynamic mode:•

Supports double display, Chinese and English Menu•

File management (support for U-disc and local storage)• Measuring speed of voltage and current: up to 500 kHz•

Built-in cold terminal compensation for thermocouple• Adjustable current rise time range: 0.001 A/us~2.5 A/us•

Comes with easy, converient and flexble any sensor measurement control

0.5 Hz• Min. readback resolution: 0.1 mV, 0.1 mA• Short-circuit, Battery test, CR-LED mode, and factory test functions

software: EasyDMM•• 4-wire SENSE compensation mode function

Standard interfaces: USB Device, USB Host, LAN (Optional Accessories: USBGPIB Adapter)• List function supports editing as many as 100 steps•

Scanner Card SC1016 (Only for SDM3065X-SC)• Program function supports 50 groups of steps•

Built-in Hlep system makes information acquisition easier• OCP, OVP, OPP, OTP and LRV protection•

Support remote control operation via SCPI commands. Compatible with• External analog control• Voltage, Current monitoring via 0-10 V• 3.5 inch TFT-LCD display, capable of displaying multiple parameters andcommands of other main stream multimeters•

Supports intelligent management system for laboratory based on BS
framework and LAN

states simultaneously• Built-in RS232/USB/LAN communication interface, USB-GPIB module(optional)• Waveform trend chart and ease-to-use file storage and call functions• Includes PC software: Supports SCPI, LabView driver

SDM3055 Series
SDM3055
SDM3055-SC (with Scanner Card)
Features and Benefits• Real 5½ digits readings resolution ( 240, 000 counts )• Up to 150 rdgs/s measurement speed• True-RMS AC Voltage and AC Current measuring• 1 Gb Nand flash size, Mass storage configuration files and data files• Built-in cold terminal compensation for thermocouple temperature
measurements• With easy, convenient and flexible PC software: EasyDMM
standard interfaces: USB Host, LAN ( Optional Accessories USB-GPIB Adapter )• Scanner Card SC1016 ( Only for SDM3055-SC )• Support remote control operation via SCPI commands.Compatible withcommands of main stream multimeters
www.siglent.com 11

Digital Multimeter

Vector Network Analyzer

SDM3045X Series

SNA5000A Series

SDM3045X
Features and Benefits• Real 4½ digit (60000 counts) readings resolution• Up to 150 rdgs/s measurement speed• True-RMS AC Voltage and AC Current measuring• 1 Gb NAND flash size, Mass storage configuration files and data files• Built-in cold terminal compensation for thermocouple• With easy, convenient and flexible PC software: EasyDMM• Standard interface: USB Device, USB Host, LAN (Optioanal Accessories:
USB-GPIB Adapter)• USB & LAN remote interfaces support common SCPI command set.
Compatible with other popular DMMs on the market

SNA5032A (100 kHz~26.5 GHz)
SNA5022A (100 kHz~13.5 GHz)
SNA5014A (9 kHz~8.5 GHz)
SNA5004A (9 kHz~4.5 GHz)
SNA5012A (9 kHz~8.5 GHz)
SNA5002A (9 kHz~4.5 GHz)
Features and Benefits:• Frequency range: 9 kHz~8.5 GHz and 100 kHz~26.5 GHz• Frequency resolution: 1 Hz
• Level resolution: 0.05 dB• Range of IFBW: 10 Hz~3 MHz• Setting range of output level: -55 dBm ~ +10 dBm• Dynamic range: 125 dB• Types of calibration: Response calibration, Enhanced Response calibration,
Full-one portcalibration, Full-two port calibration, Full-three port calibration,
Full-four port calibration, TRL calibration• Types of measurement: Scattering-parameter measurement,
differential-parameter measurement,receiver measurement, time-domain
parameter analysis, limit test, ripple test, impedance conversion, fixture
simulation, adapter removal/insertion, enhanced time-domain parameter
analysis (TDR), spectrum analysis, frequency offset, scalar mixer measurement• Support Bias-Tees• Interface: LAN, USB Device, USB Host(USB-GPIB)• Remote control: SCPI/Labview/IVI based on USB-TMC/VXI-11/Socket/Telnet/
WebServer• 12.1-inch touch screen• Video output: HDMI

www.siglent.com 12

Portable Vector Network Analyzer

Switch Matrix

SHN900A Series

SSM5000A Series

SHN926A (30 kHz~26.5 GHz)
SHN920A (30 kHz~20 GHz)
SHN914A (30 kHz~14 GHz)

SSM5342A (100 kHz~26.5 GHz)
SSM5321A (100 kHz~26.5 GHz)
SSM5144A (9 kHz~9 GHz)
SSM5142A (9 kHz~9 GHz)
SSM5124A (9 kHz~9 GHz)
SSM5122A (9 kHz~9 GHz)

Features and Benefits:• Standard VNA and CAT mode, optional SA mode
• Frequency range: 30 kHz - 26.5 GHz• Frequency resolution: 1 Hz
• Level resolution: 0.05 dB• Range of IFBW: 10 Hz~3 MHz• Setting range of output level: -45 dBm ~ +10 dBm• Dynamic range: 110 dB(Typ.)• Types of calibration: Response calibration, Enhanced Response calibration,
Full-one port calibration, Full-two port calibration, TRL calibration• Types of measurement: Scattering-parameter measurement, differentialparameter measurement, receiver measurement, time-domain parameter
analysis, limit test, ripple test, impedance conversion, fixture simulation,
adapter removal/insertion, spectrum analysis frequency offset, scalar mixer
measurement, pulse measurement

Features and Benefits:• Characteristic impedance: 50 Ω
• Highest frequency: 9 GHz (or 26.5 GHz)
• Maximum number of input ports: 4• Maximum number of output ports: 24• RF connector: 3.5 mm female• Maximum input power: 20 dBm• Maximum input DC voltage: 35 V
• Interface: LAN, USB Device, Direct Control (in), Direct Control (out) Screen
size: 2.4-inch• Internal Bias-Tee connections• Support GPS, Time and Location Information Saving• Interface: LAN, USB Device, USB Host (USB-GPIB)• Remote control: SCPI/ Labview/ IVI based on USB-TMC / VXI-11 / Socket /Telnet / WebServer• 8.4-inch touch screen, Mouse, Keyboard• Typical working time 4 hours, 3.2 kg net weight

www.siglent.com 13

Mechanical Switch

Spectrum Analyzer

SSU5000A Series

SSA5000A Series

SSU5504A (DC~50 GHz)
SSU5503A (DC~50 GHz)
SSU5502A (DC~50 GHz)
SSU5501A (DC~50 GHz)
SSU5266A (DC~26.5 GHz)
SSU5265A (DC~26.5 GHz)
SSU5264A (DC~26.5 GHz)
SSU5263A (DC~26.5 GHz)
SSU5262A (DC~26.5 GHz)
SSU5261A (DC~26.5 GHz)
SSU5184A (DC~18 GHz)
SSU5183A (DC~18 GHz)
SSU5182A (DC~18 GHz)
SSU5181A (DC~18 GHz)

SSA5085A (9 kHz~26.5 GHz)
SSA5083A (9 kHz~13.6 GHz)

Features and Benefits:• Maximum frequency: 18 GHz/ 26.5 GHz/ 50 GHz• 1 to 4 SPDT switches or 1 to 2 SP6T switch configurations• SCPI Controllable via VISA and EasySSU software• USB Connectivity• Size: WxHxD=153x62.4x137.5 mm• RF connector: SMA female or 2.4 mm female

www.siglent.com 14

Features and Benefits:• Spectrum Analyzer Frequency Range from 9 kHz up to 13.6 GHz/26.5 GHz• -165 dBm/Hz Displayed Average Noise Level (Typ.)• -105 dBc/Hz@1 GHz,10 kHz offset SSB Phase Noise (Typ.)• 25 MHz/40 MHz Analysis Bandwidth• 100% POI 7.20 μs, Dynamic Range 60 dB, Multi-view for Density, Spectrogramand PvT• Channel power, ACPR, OBW, Harmonic, TOI measurement etc.
• Analog Modulation Analysis and Vector Digital Modulation Analysis• 12.1 inch Multi-Touch Screen, HDMI output• Web Browser Remote Control on PC and Mobile Terminals and File Operation

Spectrum Analyzer

Real-Time Spectrum Analyzer

SSA3000X Plus Series

SSA3000X-R Series

SSA3075X Plus (9 kHz~7.5 GHz)
SSA3032X Plus (9 kHz~3.2 GHz)
SSA3021X Plus (9 kHz~2.1 GHz)
SSA3015X Plus (9 kHz~1.5 GHz)

SSA3075X-R (9 kHz~7.5 GHz)
SSA3050X-R (9 kHz~5.0 GHz)
SSA3032X-R (9 kHz~3.2 GHz)

Features and Benefits:• -165 dBm/Hz Displayed Average Noise Level (Typ.)Features and Benefits:• -165 dBm/Hz Displayed Average Noise Level (Typ.)• -98 dBc/Hz.@10 kHz Offset Phase Noise (1 GHz, Typ.)• -98 dBc/Hz.@10 kHz Offset Phase Noise (1 GHz, Typ.)• Level Measurement Uncertainty < 0.7 dB (Typ.)• Level Measurement Uncertainty < 0.7 dB (Typ.)• 1 Hz Minimum Resolution Bandwidth (RBW)• 1 Hz Minimum Resolution Bandwidth (RBW)• Preamplifier Standard• Preamplifier Standard• Tracking Generator Standard• Tracking Generator• Up to 40 MHz Real Time Analysis Bandwidth• Analog and Digital Signal Modulation Analysis Mode (Opt.)• 100% POI 7.20 μs, Dynamic Range 60 dB• EMI Filter and Quasi-Peak Detector Kit(Opt.)• Multi-view for Density, Spectrogram, PvT, and multi trigger and FMT• 10.1 lnch Multi-Touch Screen , Mouse and Keyboard supported• Modulation Analysis up to 40 MHz BW (Opt.)• Web Browser Remote Control on PC and Mobile Terminals and File Operation• EMI Filter and Quasi-Peak Detector Kit(Opt.)• 10.1 inch Multi-Touch Screen , Mouse and Keyboard supported• Web Browser Remote Control on PC and Mobile Terminals and File Operation

SSA3000X Series
SSA3032X (9 kHz~3.2 GHz)
SSA3021X (9 kHz~2.1 GHz)
Features and Benefits:• -161 dBm/Hz Displayed Average Noise Level (Typ.)• -98 dBc/Hz @10 kHz Offset Phase Noise (1 GHz, Typ.)• Total Amplitude Accuracy < 0.7 dB• 1 Hz Minimum Resolution Bandwidth (RBW)• Preamplifier Standard• Up to 3.2 GHz Tracking Generator (Standard)• Reflection Measurement Kit (Opt.)• Advanced Measurement Kit (Opt.)• EMI Pre-compliance Measurements Kit (Opt.)• 10.1 lnch WVGA (1024x600) Display

www.siglent.com 15

Portable Spectrum Analyzer

Spectrum & Vector Network Analyzer

SHA850A Series

SVA1000X Series

SHA852A (9 kHz~7.5 GHz)
SHA851A (9 kHz~3.6 GHz)

SVA1075X (9 kHz~7.5 GHz)
SVA1032X (9 kHz~3.2 GHz)
SVA1015X (9 kHz~1.5 GHz)

Features and Benefits:• Spectrum Analyzer Frequency Range from 9 kHz up to 7.5 GHz, -165 dBm/
Hz Displayed Average Noise Level (Typ.), -104 dBc/Hz.@10 kHz Offset Phase

Features and Benefits:• Vector Network Analyzer Frequency Range from 100 kHz up to 7.5 GHz

Noise (1 GHz, Typ.), 1 Hz up to 10 MHz Minimum Resolution Bandwidth (RBW),• -165 dBm/Hz Displayed Average Noise Level (Typ.)Preamplifier and independent signal source up to 7.5 GHz, GPS positioning and• -98 dBc/Hz.@10 kHz Offset Phase Noise (1 GHz, Typ.)logging• Level Measurement Uncertainty < 0.7 dB (Typ.)• Cable and Antenna Test Frequency Range from 100 kHz up to 7.5 GHz,
Distance To Fault and Time Domain Analysis• 1 Hz Minimum Resolution Bandwidth (RBW)• Preamplifier Standard• Vector Network Analyzer, Bias out up to 32VDC• Tracking Generator Standard• Typical working time 4 hours, 3.2 kg net weight, 8.4 lnch Multi-Touch Screen ,• Distance To Fault (Opt.)

Mouse and Keyboard supported• Analog and Digital Signal Modulation Analysis Mode (Opt.)• EMI Filter and Quasi-Peak Detector Kit(Opt.)• Advanced Measurement Kit (Opt.)• 10.1 lnch Multi-Touch Screen , Mouse and Keyboard supported• Web Browser Remote Control on PC and Mobile Terminals and File Operationwww.siglent.com 16

RF Signal Generator

RF Signal Generator

SSG6000A Series
SSG6087A (CW MODE 100 kHz ~ 40 GHz)
SSG6085A (CW MODE 100 kHz ~ 20 GHz)
SSG6083A (CW MODE 100 kHz ~ 13.6 GHz)
Features and Benefits• Frequency up to 13.6 GHz/ 20 GHz/ 40GHz• 0.01 Hz frequency setting resolution• Level setting range: -130 dBm ~ 24 dBm• Phase Noise: -135 dBc / Hz @ 1 GHz, 20 kHz offset (typ.)• Level error ≤ 0.7 dB (typ.)

SSG5000X Series
SSG5060X (CW MODE 9 kHz ~ 6 GHz)
SSG5040X (CW MODE 9 kHz ~ 4 GHz)
SSG5060X-V (CW MODE 9 kHz ~ 6 GHz /IQ MODE 10 MHz ~ 6 GHz)
SSG5040X-V (CW MODE 9 kHz ~ 4 GHz /IQ MODE 10 MHz ~ 4 GHz)

• Provides AM analog modulation with internal, external or Int+Ext source

Features and Benefits• Single pulse, double pulse and pulse train generator (option)• Frequency up to 4 GHz/6 GHz• The power meter control kit can easily use the power meter to measure• 0.001 Hz frequency setting resolution

power, control power output and correct line loss 5 inch TFT capacitive• High output power up to +26 dBm (typ.)touch screen, mouse and keyboard supported• Phase Noise: -120 dBc/ Hz @ 1 GHz, 20 kHz offset (typ.)• Web browser remote control on PC and mobile terminals• User flatness correction with power sensor to correct the cable loss

• Standard interface includes USB Host, USB Device (USB TMC), LAN (VXI-11,• Provides AM, FM, PM analog modulation with internal, external or Int+Ext

Socket, Telnet). Optional interface: GPIB
• Standard OCXO and step attenuator (0 to 110 dB, 10 dB step)source• Single pulse, double pulse and Pulse train generator (option)• Internal IQ modulation with 150 MHz modulation bandwidth with perfect infactory calibration• Internal include some digital communication stand file such as 5G-NR, LTE,WCDMA, WLAN, and playback them
• Internal Custom mode generate common IQ signal such as QAM, FSK, ASK,
MSK
• Analog differential I/Q outputs• External analog I/Q input• USB-power meter measurement• 5inch TFT capacitive touch screen, mouse and keyboard supported• Web browser remote control on PC and mobile terminals• Standard interface included USB Host, USB Device (USB TMC), LAN (VXI-11,

SSG5000A Series

Socket, Telnet). Optional interface: GPIB

SSG5085A (CW MODE 9 kHz ~ 20 GHz)
SSG5083A (CW MODE 9 kHz ~ 13.6 GHz)
Features and Benefits• Frequency up to 13.6 GHz / 20 GHz• 0.001 Hz frequency setting resolution• Level setting range: -130 dBm ~ 25 dBm• Phase Noise: -120 dBc / Hz @ 1 GHz, 20 kHz offset (typ.)• Level error ≤ 0.7 dB (typ.)• Provides AM, FM, PM analog modulation with internal, external or Int+Ext
source• Single pulse, double pulse and pulse train generator (option)• The power meter control kit can easily use the power meter to measure
power, control power output and correct line loss
• 5 inch TFT capacitive touch screen, mouse and keyboard supported• Web browser remote control on PC and mobile terminals• Standard interface includes USB Host, USB Device (USB TMC), LAN (VXI-11,
Socket, Telnet). Optional interface: GPIB
www.siglent.com 17

RF Signal Generator

SSG3000X Series
SSG3032X (CW MODE 9 kHz~3.2 GHz)
SSG3021X (CW MODE 9 kHz~2.1 GHz)
SSG3032X-IQE (IQ MODE 10 MHz~3.2 GHz)
SSG3021X-IQE (IQ MODE 10 MHz~2.1 GHz)
Features and Benefits• 0.01 Hz frequency setting resolution• Level output from -110 dBm to +13 dBm• Maximum level up to +20 dBm (typ.)• Phase Noise: -110 dBc/ Hz @ 1 GHz , 20 kHz offset (typ.)• Level accuracy ≤0.7 dB (typ.)• Provides AM, FM, &PM analog modulation with internal, external or Int+Ext
source• Pulse modulation, on/off ratio ≥70 dBc• Pulse train generator (option)• External IQ modulation with SDG6000X as the baseband IQ signal• USB-power meter measurement• 5 inch TFT capacitive touch screen, mouse and keyboard supported• Web browser remote control on PC and mobile terminals• Standard interface include USB Host, USB Device (USB TMC), LAN (VXI-11,
Socket, Telnet). Optional interface: GPIB

www.siglent.com 18

Probes and Accessories
Type

Model

PictureSpecifications

PB470
PP510
PP215

PB470, 70 MHz bandwidth
PP510, 100 MHz bandwidth
PP215, 200 MHz bandwidth
1 X/10 X decay, 1 M/10 Mohm, 300 V/600 V

PB925

10X CAT II 1000 V, CAT III 600 V, 250 MHz-bandwidth, SHS1000 handheld
oscilloscope option

SAP1000

Active Probe, 1 GHz

SAP2500

Active Probe, 2.5 GHz

SAP2500D

Differential Active Probe, 2.5 GHz

SAP5000D

Differential Active Probe, 5 GHz

CPL5100

Bandwidth: DC-600 kHz ; Current Range: L (50 mA~10 A Peak),H(1 A~100 A Peak); Attenuation accuracy L (0.1 V/A), H (0.01 V/A); Typical
DC precision: L (3%±50 mA), H(500 mA~40 A Peak : 4%±50 mA; 4 0A~100
A Peak : ±15% Maximum); Rise Time: ≤583 ns; Operating voltage RMS:
CATI 600 V CATII 600 V CATIII 300 V; 9 V alkaline layer-built battery/ 15H

CP4020

Bandwidth: 100 kHz; Maximum continuous current 20 Arms; Peak current 60
A; Switching ratio: 50 m /A; 5 mV/A;
DC measurement accuracy: 50 mV/A (0.4 A-10 ApK) ± 2%; 5 mV/A
(1 A-60 ApK)±2%; 9 V battery-powered

CP4050

Bandwidth: 1 MHz; Maximum continuous current 50 Arms;
Peak current 140 A; Switching ratio: 500 mV/A; 50 mV/A;
DC measurement accuracy: 500 mV/A (20 mA-14 ApK) ±3%±20 mA;
50 mV/A (200 mA-100 ApK )±4%± 200 mA;
50 mV/A (100 A-140 ApK)±15% max; 9 V battery-powered

CP4070

Bandwidth: 150 KHz; Maximum continuous current 70 Arms; Peak current
200 A; Switching ratio: 50 mV/A; 5 mV/A;
DC measurement accuracy: 50 mV/A(0.4A-10ApK)±2%; 5 mV/A
(1 A-200 ApK)±2%; 9 V battery-powered

CP4070A

Bandwidth: 300 kHz; Maximum continuous current 70 Arms; Peak current
200 A;Switching ratio：100 mV/A;10 mV/A;
DC measurement accuracy：100 mV/A (50 mA-10 ApK) ±3%±50 mA;
10 mV/A (500 mA-40 ApK) ±4%±50 mA;
10 mV/A (40 A-200 ApK) ±15% max; 9 V battery-powered

Passive Probe

Active Probe

Current Probewww.siglent.com 19

Type

Model

USB AWG
ModuleSAG1021I

Specifications
Frequency is determined by oscilloscope. Isolated USB function/arbitrary
waveform generator, 125 Msa/s, 16 kpts Arb Wave Length, Insulation Voltage,±42 Vpk(Hardware)

BAG-H2Carry Bag

PictureBAG-S1

BAG-S2

Soft Carry Case for SHA850A, SHN900A

Soft Carry Case for SDS1000DL+/CML+, SDS1000X, SDS1000X-E, SDS2000X-E
SeriesSoft Carry Case for SDS2000X, SDS2000X Plus, SDS2000X HD, SDS5000X,
SSA3000X, SSA3000X Plus, SVA1000X, SSA3000X-R

Bandwidth: 50 MHz; Maximum continuous current 30 Arms; Peak current 50 A;
SCP5030

Switching ratio: 5 A/30 A;
Accuracy: 5 A(±1%±1 mA); 30 A(±1%±10 mA); Powered by oscilloscope via
SAPBUS
Bandwidth: 100 MHz; Maximum continuous current 30 Arms; Peak current 50

SCP5030A

A; Switching ratio: 5 A/30 A;
Accuracy: 5 A(±1%±1 mA); 30 A(±1%±10 mA); Powered by oscilloscope via
SAPBUS
Bandwidth: 12 MHz; Maximum continuous current 150 Arms; Peak current

SCP5150300 A; Switching ratio: 30 A/150 A;
Accuracy: 30 A(±1%±10 mA); 150 A(±1%±100 mA); Powered by oscilloscope
via SAPBUS
Bandwidth: 2 MHz; Maximum continuous current 500 Arms; Peak current 750

SCP5500

A; Switching ratio: 75 A/500 A;
Accuracy: 75 A(±1%±10 mA); 500 A(±1%±100 mA); Powered by oscilloscope
via SAPBUS

Current Probe
Bandwidth: 50 MHz; Maximum continuous current 30 Arms; Peak current 50 A;
CP6030Switching ratio: 5 A/30 A;
Accuracy: 5 A(±1%±1 mA); 30 A(±1%±10 mA); Standard DC12 V/1 A power
adapter
Bandwidth: 100 MHz; Maximum continuous current 30 Arms; Peak current 50

CP6030A

A; Switching ratio: 5 A/30 A;
Accuracy: 5 A(±1%±1 mA); 30 A(±1%±10 mA); Standard DC12 V/1 A power
adapter
Bandwidth: 12 MHz; Maximum continuous current 150 Arms; Peak current

CP6150300 A; Switching ratio: 30 A/150 A;
Accuracy: 30 A(±1%±10 mA); 150 A(±1%±100 mA); Standard DC12 V/1 A
power adapter
Bandwidth: 5 MHz; Maximum continuous current 500 Arms; Peak current 750

CP6500A; Switching ratio: 75 A/500 A;
Accuracy: 75 A(±1%±10 mA); 500 A(±1%±100 mA); Standard DC12 V/1 A
power adapter

www.siglent.com 20

Type

Model

PictureSpecifications
Bandwidth: 50 MHz; Maximum continuous current 30 Arms; Peak current 50 A;

CP6030Switching ratio: 5 A/30 A;
Accuracy: 5 A(±1%±1 mA); 30 A(±1%±10 mA); Standard DC12 V/1 A power
adapter
Bandwidth: 100 MHz; Maximum continuous current 30 Arms; Peak current 50

CP6030A

A; Switching ratio: 5 A/30 A;
Accuracy: 5 A(±1%±1 mA); 30 A(±1%±10 mA); Standard DC12 V/1 A power
adapterCurrent Probe
Bandwidth: 12 MHz; Maximum continuous current 150 Arms; Peak current
CP6150300 A; Switching ratio: 30 A/150 A;
Accuracy: 30 A(±1%±10 mA); 150 A(±1%±100 mA); Standard DC12 V/1 A
power adapter
Bandwidth: 5 MHz; Maximum continuous current 500 Arms; Peak current 750

CP6500A; Switching ratio: 75 A/500 A;
Accuracy: 75 A(±1%±10 mA); 500 A(±1%±100 mA); Standard DC12 V/1 A
power adapter
Bandwidth: 50 MHz; Rise Time ≤7 ns; DC Accuracy ±2%; Max Input: 600 V

DPB1300

CATIII, 1000 V CATII; Max Differential Test Voltage (DC + Peak AC): 50 X:±130 V, 500 X: ±1300 V. Input impedance/capacitance: 5 MΩ/<4 pF(Singleended), 10 MΩ/< 2 pF(Two inputs); DC 12 V/1.2 A Power

Bandwidth: 50 MHz; Maximum input differential voltage 800 V (DC + Peak
DPB4080AC); Range selection (attenuation ratio):10 X/100 X; Accuracy: ±1%;
Standard DC 9 V/1 A power adapter

Bandwidth: 70 MHz; Maximum input differential voltage 1500 V (DC +DPB5150

Peak AC); Range selection (attenuation ratio): 50 X/500 X; Accuracy: ±2%;Standard 5 V/1 A USB power adapter

High Voltage
Differential
Probe
Bandwidth: 100 MHz; Maximum input differential voltage 1500 V (DC +DPB5150A

Peak AC); Range selection (attenuation ratio): 50 X/500 X; Accuracy: ±2%;Standard 5 V/1 A USB power adapter

Bandwidth: 70 MHz; Maximum input differential voltage 7000 V (DC + Peak
DPB5700

AC); Range selection (attenuation ratio): 100 X/1000 X; Accuracy: ±2%;Standard 5 V/1 A USB power adapter

Bandwidth: 100 MHz; Maximum input differential voltage 7000 V (DC + Peak
DPB5700AAC);Range selection (attenuation ratio): 100 X/1000 X; Accuracy: ±2%;Standard 5 V/1 A USB power adapter

High Voltage
Probe

HPB4010

Logic Probe

SPL2016

Bandwidth: 40 MHz; Maximum input differential voltage DC: 10 kV; AC(rms):
7 kV(sine); AC(Vpp): 20 kV(Pulse); attenuation ratio1: 1000; Accuracy: ≤3%

Logic Probe, 16-channel, 500 MSa/s

www.siglent.com 21

Type

Model

Logic Analyzer

SLA1016

PictureSpecifications
MSO function hardware for SDS2000X-E oscilloscope, 16-channel, 500 MSa/s,
14 Mpts

Three magnetic field near-field probes and one electric field near field probe;
Near-Field
Probe

SRF5030T

Frequency range: 30 MHz~3 GHz; resolution 25 mm; distinguished within
10 cm range of the magnetic field; for EMI radiation interference and theintensity detector

Deskew Fixture

Cablewww.siglent.com 22

DF2001A

Deskew fixture for voltage and current probes

N-BNC-2L

N-BNC cable, 2 GHz bandwidth

N-N-6L

N-N cable, 6 GHz bandwidth

N-SMA-6L

N-SMA cable, 6 GHz bandwidthN-N-18L

N(M)-N(M) cable, 18 GHz

N-SMA-18L

N(M)-SMA(M) cable, 18 GHz

SMA-SMA-18L

SMA(M)-SMA(M) cable, 18 GHz

SMA-SMA-26L

SMA(M)-SMA(M) cable, 26 GHz

SMAF-SMA-26L

SMA(F)-SMA(M) cable, 26 GHz

2.92F-2.92F-40A

2.92 mm Female - 2.92 mm Female adaptor, 40 GHz

V26-N35MN35F-25IN

NMD 3.5 mm male - NMD 3.5 mm female; 26.5 GHz; length 25''/635 mm

V26-N35FA35F-25IN

NMD 3.5 mm female - APC 3.5 mm female; 26.5 GHz; length 25''/635 mm

Type

Model

GPIB

USB-GPIB

Isolated Front
End

ISFE

STB Test board

STB-3

For experimental teaching and product demos

FX-USB2

USB 2.0 test fixture

FX-ETH

100Base-TX & 1000Base-T compliance test fixture

FX-AMETH

100Base-T1 & 1000Base-T1 compliance test fixture

Test fixture

PictureSpecifications

USB-GPIB Adapter, USB Device expanded into GPIB interface

USB 5V power supply, plug and play, the maximum input voltage 600Vp-p,
floating test. Work with oscilloscopes

Rackmount kit for two intruments ,SDG800, SDG1000X, SDG2000X,
SDG-2-RMK

SDG5000,SDG6000X series generators and SDM digital multimeters;
Height 3U

SDS1X-E-RMK

Rackmount kit , compatible with the SDS800X HD, SDS1000X-E, SDS1000X-U,
SDS2000X-E model; Height 4U

Rackmount kit, compatible with SDG800, SDG1000X, SDG2000X, SDG5000,
SDG-RMK

SDG6000X series generators and SDM digital multimeters, SDL1000X load;
Height 3U

Rack Mount

SSA-RMKSSG-RMKSDS2000 HD-RMK

Rackmount kit , compatible with the SSA3000X, SSA3000X Plus, SVA1000X,
SSA3000X-R model; Height 6U

Rack Mount kit; SSG3000X, SSG5000X, SSG5000A, SDG7000A; Height 3U

Rack Mount kit for SDS1000X HD, SDS2000X HD, SDS3000X HD; Height
6U(exactly 260 mm)

www.siglent.com 23

Type

Model

PictureSpecifications

SPD3000-RMK

Rackmount kit , compatible with the SPD3000X/X-E/D/S/C models, Height 4U

SPS5000X-RMK

SPS5000X EIA Standard rack, Height 3U

SDS2000-RMK

Rackmount kit is designed for use with only one instrument, is compatible
with the SDS2000,SDS2000X, SDS2000X Plus series Oscilloscope; Height 6U

Rack Mount

SDS5000X-RMKRack Mount kit for SDS5000X; Height 6U

SDS6000-RMK

Rack Mount kit for SDS6000A, SNA5000A, SSA5000A; Height 7U

SSG6000A-RMKRack Mount kit; SSG6000A; Height 2U

F503ME

4.5 GHz

F503FE

Mechanical Calibration Kit: OSLT, DC - 4.5 GHz, N-Female connector

F504MS

Mechanical Calibration Kit: OSLT, DC - 9 GHz, N-Male connector

F504FS

Mechanical Calibration Kit: OSLT, DC - 9 GHz, N-Female connector

F504TS

Mechanical Calibration Kit: OSLT, DC - 9 GHz, N-Male and Female connector

F505TS

Mechanical Calibration Kit: OSLT, DC - 18 GHz, N-Male and Female connector

VNA Calibration
Kit
F603ME

F603FE

F604MS

F604FS

F604TS

F604TYwww.siglent.com 24

Mechanical Calibration Kit: Open (M), Short (M), Match (M,50), Through (F-F),

Mechanical Calibration Kit: OSLT, DC - 4.5 GHz, 3.5 mm SMA-Maleconnector
Mechanical Calibration Kit: Open (M), Short (M), Match (M,50), Through (F-F),
4.5 GHz, SMA-typeMechanical Calibration Kit: OSLT, DC - 9 GHz, 3.5 mm SMA-Maleconnector
Mechanical Calibration Kit: OSLT, DC - 9 GHz, 3.5 mm SMA-Female
connector
Mechanical Calibration Kit: OSLT, DC - 9 GHz, 3.5 mm-Male and Female
connector
Mechanical Calibration Kit: OSLT, DC - 27 GHz, 3.5 mm-Male and Female
connector

Type

Model

PictureSpecifications

Y504MS

Integrated Mechanical Calibration Kit: OSLT, DC - 9 GHz, N-Male

Y504FS

Integrated Mechanical Calibration Kit: OSLT, DC - 9 GHz, N-Female

VNA Calibration
Kit

Increase the voltage and current output capabilities to generators like the
SIGLENT SDG family.
Typical Input Impedance: 15 kΩ
Input:+/- 6.5 V Vpp (Gain: X1)
Amplifier

SPA1010+/- 1.3 V (Gain: X10)
Gain: Switched 10 V/1 V and 10 V/10 V
Output Voltage: 25.4 Vpp
Output Current: 1.12 A
Slew Rate: ≥ 90 V/μs
Overshoot: ≤ 4%
Compatible with all SIGLENT SDG series generators

Synchronous
ModuleSYN64

64 synchronous module

ANT-GPS1

GPS antenna, SMA(M), 100 cm

Directional Antenna Suit, N type,
ANT-DA11 antenna (10 MHz~200 MHz),ANT-DA1

ANT-DA12 antenna (200 MHz~500 MHz),ANT-DA13 antenna (500 MHz~8 GHz),
Amplifier handle 12dB@1GHz(typ.)Contains amplifier handle and 10 MHz ~ 200 MHz antenna. Antenna gain
Antenna

ANT-DA11

10 dB (typical value); SWR <1:1.9 (typical value); 50 Ω/N type, female;polarization direction horizontal and verticalContains amplifier handle and 200 MHz ~ 500 MHz antenna. Antenna gain
ANT-DA12

10 dB (typical value); SWR <1:1.9 (typical value); 50 Ω/N type, female;polarization direction horizontal and verticalContains amplifier handle and 500 MHz ~ 8 GHz antenna. Antenna gain 10 dB
ANT-DA13(typical value); SWR <1:1.9 (typical value); 50 Ω/N type, female; polarization
direction horizontal and vertical

RF Test board

SNA-TB01

Board integrated with RF components like amplier, mixer, filter for vector
network analyzer demonstrationwww.siglent.com 25

Type

Model

PictureSpecifications

ADP-18Adjustable differential TDR probe DC~18 GHz

ADP-26

Adjustable differential TDR probe DC~26.5 GHz

ASP-18Adjustable single-end TDR probe DC~18 GHz

ASP-26

Adjustable single-end TDR probe DC~26.5 GHz

SSM5122A

2 input ports, 12 output ports, 3.5 mm female, 9 kHz~9 GHz

SSM5124A

2 input ports, 24 output ports, 3.5 mm female, 9 kHz~9 GHz

SSM5142A

4 input ports, 12 output ports, 3.5 mm female, 9 kHz~9 GHz

SSM5144A

4 input ports, 24 output ports, 3.5 mm female, 9 kHz~9 GHz

SSM5321A

2 input ports, 6 output ports, 3.5 mm female, 100 kHz~26.5 GHz

SSM5342A

4 input ports, 12 output ports, 3.5 mm female, 100 kHz~26.5 GHz

TDR Probe

Switch Matrix

Utility Kit for SSA3000X Series: N (M) -SMA (M) cable, N (M) -N (M) cable, N (M)

SSA3000X
Utility Kit

UKitSSA3X

Rechargeable
lithium battery

10V8_BAT

10.8V, 74 Wh

AC-DC adapter

12V_AP_4A

12V, 4A

Reflection
BridgeRB3X25

PC Software

SigIQPro

Noise Source
Driver

NSD28

www.siglent.com 26-BNC (F) adaptor (2 pcs), N (M) -SMA (F) adaptor (2 pcs), 10 dB attenuator;

RB (1 MHz~2.5 GHz), N (M) -N (M) adaptor (2 pcs), for SSA3000X,
SSA3000X Plus series

A comprehensive PC-based software for general and standards-based signals
creation, supporting Bluetooth, IoT, Custom OFDM, etc.

Noise source driver, connect spectrum analyzer to noise source

Type

Model

PictureSpecifications

SEM5002A

2 ports, 9 kHz~4.5 GHz, SMA female

SEM5012A

2 ports, 9 kHz~9 GHz, SMA female

SEM5022A

2 ports, 100 kHz~13.5 GHz, 3.5 mm female

SEM5032A

2 ports, 100 kHz~26.5 GHz, 3.5 mm female

SEM5004A

4 ports, 9 kHz~4.5 GHz, SMA female

SEM5014A

4 ports, 9 kHz~9 GHz, SMA female

SEM5024A

4 ports, 100 kHz~13.5 GHz, 3.5 mm female

SEM5034A

4 ports, 100 kHz~26.5 GHz, 3.5 mm female

SSU5181A

DC~18 GHz, including one SPDT mechanical switch, SMA female

SSU5182A

DC~18 GHz, including two SPDT mechanical switchs, SMA female

SSU5183A

DC~18 GHz, including three SPDT mechanical switchs, SMA female

SSU5184A

DC~18 GHz, including four SPDT mechanical switchs, SMA female

SSU5261A

DC~26.5 GHz, including one SPDT mechanical switch, SMA female

SSU5262A

DC~26.5 GHz, including two SPDT mechanical switchs, SMA female

SSU5263A

DC~26.5 GHz, including three SPDT mechanical switchs, SMA female

SSU5264A

DC~26.5 GHz, including four SPDT mechanical switchs, SMA female

SSU5265A

DC~26.5 GHz, including one SP6T mechanical switch, SMA female

SSU5266A

DC~26.5 GHz, including two SP6T mechanical switchs, SMA female

SSU5501A

DC~50 GHz, including one SPDT mechanical switch, 2.4 mm female

SSU5502A

DC~50 GHz, including two SPDT mechanical switchs, 2.4 mm female

SSU5503A

DC~50 GHz, including three SPDT mechanical switchs, 2.4 mm female

SSU5504A

DC~50 GHz, including four SPDT mechanical switchs, 2.4 mm female

VNA Calibration
Kit

Mechanical
Switchwww.siglent.com 27

About SIGLENTSIGLENT is an international high-tech company, concentrating on R&D, sales,
production and services of electronic test & measurement instruments.
SIGLENT first began developing digital oscilloscopes independently in 2002. After
more than a decade of continuous development, SIGLENT has extended its productline to include digital oscilloscopes, isolated handheld oscilloscopes, function/arbitrary
waveform generators, RF/MW signal generators, spectrum analyzers, vector network
analyzers, digital multimeters, DC power supplies, electronic loads and other general
purpose test instrumentation. Since its first oscilloscope was launched in 2005,
SIGLENT has become the fastest growing manufacturer of digital oscilloscopes. We
firmly believe that today SIGLENT is the best value in electronic test & measurement.

Headquarters:
SIGLENT Technologies Co., Ltd
Add: Bldg No.4 & No.5, Antongda Industrial
Zone, 3rd Liuxian Road, Bao'an District,
Shenzhen, 518101, China
Tel: + 86 755 3688 7876
Fax: + 86 755 3359 1582
Email: sales@siglent.com
Website: int.siglent.com
USA:
SIGLENT Technologies America, Inc
6557 Cochran Rd Solon, Ohio 44139
Tel: 440-398-5800
Toll Free: 877-515-5551
Fax: 440-399-1211
Email: info@siglent.com
Website: www.siglentna.com
Europe:SIGLENT Technologies Germany GmbH
Add: Staetzlinger Str. 70
86165 Augsburg, Germany
Tel: +49(0)-821-666 0 111 0
Fax: +49(0)-821-666 0 111 22
Email: info-eu@siglent.com
Website: www.siglenteu.com

Follow us on
Facebook: SiglentTech

modiﬁcation date：2024-05