SIGLENT TECHNOLOGIESRF PRODUCT CATALOG· Spectrum Analyzer· Spectrum & Vector Network Analyzer· Vector Network Analyzer · RF/MW Signal Generator

Catalog
Company Profile .................................................................................................................................................................................. 02
Vector Network Analyzer

SNA5000A Vector Network Analyzer .......................................................................................................................................... 03
SHN900A Portable Vector Network Analyzer ........................................................................................................................ 08

Switch Matrix

SSM5000A Switch Matrix .................................................................................................................................................................... 12

Mechanical Switch

SSU5000A Mechanical Switch ........................................................................................................................................................ 14

Spectrum Analyzer

SSA5000A Spectrum Analyzer ......................................................................................................................................................... 16
SSA3000X-R Real-time Spectrum Analyzer ............................................................................................................................. 19
SSA3000X Plus Spectrum Analyzer .............................................................................................................................................. 23
SSA3000X Spectrum Analyzer .......................................................................................................................................................... 26
SHA850A Portable Spectrum Analyzer ...................................................................................................................................... 29

Spectrum & Vector Network Analyzer

SVA1000X Spectrum & Vector Network Analyzer ............................................................................................................. 32

RF/MW Signal Generator

SSG6000A RF Signal Generator ...................................................................................................................................................... 36
SSG5000A RF Signal Generator ...................................................................................................................................................... 39
SSG5000X RF Signal Generator ........................................................................................................................................................ 41

SSG3000X RF Signal Generator ........................................................................................................................................................ 46

Accessories ............................................................................................................................................................................................... 50
Other Products Overview ........................................................................................................................................................... 55
Service .......................................................................................................................................................................................................... 59

www.siglent.com 1

Company Profile

SIGLENT TECHNOLOGIES Co., Ltd.
Every Bench. Every Engineer. Every Day.
SIGLENT has been providing test & measurement solutions for almost 18 years from its headquarterin Shenzhen, China. There are more than 300 employees, one third of whom are high-educated R&D
engineers.
SIGLENT has many patent technologies. We are dedicated to develop sophisticated and high quality
digital oscilloscopes, waveform generators, RF signal generators, handheld digital oscilloscopes, spectrum
analyzers, vector network analyzers and DC power supplies, DC Electronic Loads, digital multimeters. We
strive to deliver the highest quality of customer service and satisfaction to our customers.

SIGLENT provides the following
instruments:-Digital Oscilloscope-Handheld Oscilloscope-Waveform Generator-RF/MW Signal Generator-Spectrum Analyzer-Vector Network Analyzer-DC Power Supply-DC Electronic Load
-Digital Multimeter-Probes & Accessories

SIGLENT sincerely invite
you to join

Please email :sales@siglent.com

www.siglent.com 2

SNA5000A
Vector Network
Analyzer

Features and Benefits• Frequency range: 9 kHz~8.5 GHz and 100 kHz~26.5 GHz• Frequency resolution: 1 Hz
• Level resolution: 0.05 dB• Range of IFBW: 10 Hz~3 MHz• Setting range of output level: -55 dBm~+10 dBm• Dynamic range: 125 dB• Types of calibration: Response calibration, Enhanced Response calibration, Full-one portcalibration, Full-two port calibration, Full-three port
calibration, Full-four port calibration, TRL calibration• Types of measurement: Scattering-parameter measurement, differential-parameter measurement, receiver measurement, time-domain
parameter analysis, limit test, ripple test, impedance conversion, fixture simulation, adapter removal/insertion, spectrum analysis frequency
offset, scalar mixer measurement• Support Bias-Tees• Interface: LAN, USB Device, USB Host(USB-GPIB)• Remote control: SCPI/Labview/IVI based on USB-TMC/VXI-11/Socket/Telnet/WebServer• 12.1-inch touch screen• Video output: HDMI

www.siglent.com 3

Design features

SNA5000A Vector Network Analyzer• Multi-window display• Multi-format display• Display and compare memory and current data

• Display data hold

• Impedance conversion• Equation Editor

www.siglent.com 4

SNA5000A Vector Network Analyzer• Port Extensions• Embedding and De-Embedding• Time-Domain analysis• Enhanced Time-Domain analysis(TDR)• Spectrum analysis• Scalar mixer measurement

www.siglent.com 5

SNA5000A Vector Network Analyzer

Model and Main index
Model

SNA5002A
SNA5004A

SNA5012A
SNA5014A

SNA5022A

SNA5032A

Frequency range

9 kHz~4.5 GHz

9 kHz~8.5 GHz

100 kHz~13.5 GHz

100 kHz~26.5 GHz

2/4

2

2

Ports

2/4

Frequency resolution

1 Hz

Level resolution

0.05 dB

Range of IFBW

10 Hz~3 MHz

Setting range of output level-55 dBm ~ +10 dBm

Dynamic range

125 dB

Types of calibration

Response calibration, Enhanced Response calibration, Full-one port calibration, Full-two port calibration, Full-three port calibration,
Full-four port calibration, TRL calibration

Types of measurement

Scattering-parameter measurement, differential-parameter measurement, receiver measurement, time-domain parameter
analysis, limit test, ripple test, impedance conversion, fixture simulation, adapter removal / insertion, enhanced time-domain
parameter analysis (TDR), spectrum analysis, frequency offset, scalar mixer measurement

Bias-Tees

SupportInterfaceLAN, USB Device, USB Host(USB-GPIB)

Remote control

SCPI/Labview/IVI based on USB-TMC/VXI-11/Socket/Telnet/WebServer

Display

12.1-inch touch screen

Video output

HDMI

Ordering Information
Items

Products

Standard Accessories

Optional Accessories

www.siglent.com 6

Description

Order Number

2 ports, 4.5G Vector Network Analyzer

SNA5002A

2 ports, 8.5G Vector Network Analyzer

SNA5012A

4 ports, 4.5G Vector Network Analyzer

SNA5004A

4 ports, 8.5G Vector Network Analyzer

SNA5014A

2 ports, 13.5G Vector Network Analyzer

SNA5022A

2 ports, 26.5G Vector Network Analyzer

SNA5032A

One Quick-start, One Power-cable, One USB-cable, One Calibration-certificate
High-performance reference source

SNA5000-HPR

Time-Domain analysis

SNA5000-TDA

Enhanced Time-Domain analysis

SNA5000-TDR

Spectrum analysis

SNA5000-SA

Scalar mixer measurement

SNA5000-SMM

N-type, Male, 50Ω Calibration Kit, 0~4.5 GHz

F503ME

N-type, Female, 50Ω Calibration Kit, 0~4.5 GHz

F503FE

3.5 mm, Male, 50Ω Calibration Kit, 0~4.5 GHz

F603ME

3.5 mm, Female, 50Ω Calibration Kit, 0~4.5 GHz

F603FE

N-type, Male, 50Ω Calibration Kit, 0~9 GHz

F504MS

N-type, Female, 50Ω Calibration Kit, 0~9 GHz

F504FSSNA5000A Vector Network Analyzer
Items

Optional Accessories

VNA Options

Description

Order Number

N-type, Male and Female, 50Ω Calibration Kit,0~9 GHz

F504TS

3.5 mm, Male, 50Ω Calibration Kit, 0~9 GH

F604MS

3.5 mm, Female, 50Ω Calibration Kit, 0~9 GHz

F604FS

3.5 mm, Male and Female, 50Ω Calibration Kit, 0~9 GHz

F604TS

3.5 mm, Male and Female, 50Ω Calibration Kit, 0~27 GHz

F604TY

N(M)-SMA(M) RF Cable DC~18 GHz,1000 mm

N-SMA-18L

N(M)-N(M) RF Cable DC~18 GHz,1000 mm

N-N-18L

SMA(M)-SMA(M) RF Cable DC~18 GHz,1000 mm

SMA-SMA-18L

SMA(M)-SMA(M) RF Cable DC~26.5 GHz,1000 mm

SMA-SMA-26L

SMA(F)-SMA(M) RF Cable DC~26.5 GHz,1000 mm

SMAF-SMA-26L

NMD 3.5 female-NMD 3.5 Male DC~26.5 GHz,635 mm

V26-N35MN35F-25IN

NMD 3.5 female-APC 3.5 female DC~26.5 GHz,635 mm

V26-N35FA35F-25IN

USB-GPIB Adapter

USB-GPIB

2 ports, 9 kHz ~ 4.5 GHz, SMA female

SEM5002A

2 ports, 9 kHz ~ 9 GHz, SMA female

SEM5012A

2 ports, 100 kHz ~ 13.5 GHz, 3.5 mm female

SEM5022A

2 ports, 100 kHz ~ 26.5 GHz, 3.5 mm female

SEM5032A

4 ports, 9 kHz ~ 4.5 GHz, SMA female

SEM5004A

4 ports, 9 kHz ~ 9 GHz, SMA female

SEM5014A

4 ports, 100 kHz ~ 13.5 GHz, 3.5 mm female

SEM5024A

4 ports, 100 kHz ~ 26.5 GHz, 3.5 mm female

SEM5034A

www.siglent.com 7

SHN900A

Portable Vector
Network Analyzer

Features and Benefits• Standard VNA and CAT mode, Optional SA mode
• Frequency range: 30 kHz - 26.5 GHz• Frequency resolution: 1 Hz
• Level resolution: 0.05 dB• Range of IFBW: 10 Hz~3 MHz• Setting range of output level: -45 dBm ~ +10 dBm• Dynamic range: 110 dB(Typ.)• Types of calibration: Response calibration, Enhanced Response calibration, Full-one port calibration, Full-two port calibration, TRL calibration• Types of measurement: Scattering-parameter measurement, differential-parameter measurement, receiver measurement, time-domain
parameter analysis, limit test, ripple test, impedance conversion, fixture simulation, adapter removal/insertion, spectrum analysis frequency
offset, scalar mixer measurement, pulse measurement• Internal Bias-Tee connections• Support GPS, Time and Location Information Saving• Interface: LAN, USB Device, USB Host (USB-GPIB)• Remote control: SCPI/ Labview/ IVI based on USB-TMC / VXI-11 / Socket /Telnet / WebServer• 8.4-inch touch screen, Mouse, Keyboard• Typical working time 4 hours, 3.2 kg net weight

www.siglent.com 8

SHN900A Portable Vector Network Analyzer
Design features• Multi-window display• Multi-format display• Impedance conversion• Equation Editor• Port Extensions• Embedding and De-Embedding

www.siglent.com 9

SHN900A Portable Vector Network Analyzer• CAT• Time-Domain analysis• Enhanced Time-Domain analysis (TDR)• Spectrum analysis

Model and Main index
Model

SHN914A

SHN920A

SHN926A

Frequency range

30 kHz~14 GHz

30 kHz~20 GHz

30 kHz~26.5 GHz

Ports

2

Frequency resolution

1 Hz

Level resolution

0.05 dB

Range of IFBW

10 Hz~3 MHz

Setting range of output level-45 dBm ~ +10 dBm

Dynamic range

110 dB (Typ.)Types of calibration

Response calibration, Enhanced Response calibration, Full-one port calibration, Full-two port calibration, TRL calibration

Types of measurement

Scattering-parameter measurement, differential-parameter measurement, receiver measurement, time-domain parameter
analysis, limit test, ripple test, impedance conversion, fixture simulation, adapter removal/insertion, enhanced timedomain parameter analysis (TDR), spectrum analysis, frequency offset, scalar mixer measurement, pulse measurement

Bias-Tees

SupportInterfaceLAN, USB Device, USB Host (USB-GPIB)

Remote control

SCPI/ Labview/ IVI based on USB-TMC/ VXI-11/ Socket/ Telnet/ WebServer

Display

8.4-inch touch screen

GPS

Supportwww.siglent.com 10

SHN900A Portable Vector Network Analyzer
Ordering Information
Items
Products

Description

Order number

2 ports, 14 G Vector Network Analyzer

SHN914A

2 ports, 20 G Vector Network Analyzer

SHN920A

2 ports, 26.5 G Vector Network Analyzer

SHN926A

standard fittings

Quick Start, USB Type C Line, Rechargeable lithium battery, AC-DC adapter, Portable bag

TDA Option

Time Domain Analysis

SHN900-TDA

TDR Option

Enhanced Time Domain Analysis

SHN900-TDR

SA Option

Spectrum analysis

SHN900-SA

3.5 mm, Male, 50 Ω Calibration Kit, 0~4.5 GHz

F603ME

3.5 mm, Female, 50 Ω Calibration Kit, 0~4.5 GHz

F603FE

3.5 mm, Male, 50 Ω Calibration Kit, 0~9 GHz

F604MS

3.5 mm, Female, 50 Ω Calibration Kit, 0~9 GHz

F604FS

3.5 mm, Male and Female, 50 Ω Calibration Kit, 0~9 GHz

F604TS

3.5 mm, Male and Female, 50 Ω Calibration Kit, 0~26.5 GHz

F606TSElectronic Calibration Kit

SEM5000A

RF Test Demo Board

SNA-TB01

Adjustable Differential TDR probe DC-18 GHz

ADP-18Adjustable Differential TDR probe DC-26.5 GHz

ADP-26

Adjustable Differential TDR probe DC-18 GHz

ASP-18Adjustable Differential TDR probe DC-26.5 GHz

ASP-26

SMA(M)-SMA(M) cable DC-18 GHz, 1000 mm

SMA-SMA-18L

SMA(M)-SMA(M) cable DC-26.5 GHz, 1000 mm

SMA-SMA-26L

SMA(F)-SMA(M) cable DC-26.5 GHz, 1000 mm

SMAF-SMA-26L

NMD 3.5 female-NMD 3.5 Male DC-26.5 GHz, 635 mm

V26-N35MN35F-25IN

NMD 3.5 female-APC 3.5 female DC-26.5 GHz, 635 mm

V26-N35FA35F-25IN

USB-GPIB Adaptor

USB-GPIB

GPS antenna, SMA(M), 1000 mm

ANT-GPS1

www.siglent.com 11

SSM5000A
Switch Matrix

Features and Benefits• Characteristic impedance: 50 Ω
• Highest frequency: 9 GHz (or 26.5 GHz)
• Maximum number of input ports: 4• Maximum number of output ports: 24• RF connector: 3.5 mm female• Maximum input power: 20 dBm• Maximum input DC voltage: 35 V
• Interface: LAN, USB Device, Direct Control (in), Direct Control (out)• Screen size: 2.4-inch

Design features• Network Analyzer Test Port Extensionwww.siglent.com 12• Signal Generator and SA Test Port Extension

SSM5000A Switch Matrix
Model and Main index
Model

SSM5122A

SSM5124A

SSM5142A

SSM5144A

SSM5321A

SSM5342A

Frequency range

9 kHz ~ 9 GHz

9 kHz ~ 9 GHz

9 kHz ~ 9 GHz

9 kHz ~ 9 GHz

100 kHz ~ 26.5 GHz 100 kHz ~ 26.5 GHz

Input

2

2

4

4

2

4

Output

12

24

12

24

6

12

RF connector

3.5 mm Female

Maximum input power

20 dBmMaximum input DC voltage

35 V

Screen size

2.4-inch

Size

W×H×D = 88.5×425×417.6 mm

Ordering Information
Items

Products

Standard Accessories

Description

Frequency range

Order number

2 input ports, 12 output ports

9 kHz ~ 9 GHz

SSM5122A

2 input ports, 24 output ports

9 kHz ~ 9 GHz

SSM5124A

4 input ports, 12 output ports

9 kHz ~ 9 GHz

SSM5142A

4 input ports, 24 output ports

9 kHz ~ 9 GHz

SSM5144A

2 input ports, 6 output ports

100 kHz ~ 26.5 GHz

SSM5321A

4 input ports, 12 output ports

100 kHz ~ 26.5 GHz

SSM5342A

One Quick-start, One Power-cable,
One USB-cable, One certificate of qualification--www.siglent.com 13

SSU5000A

Mechanical Switch

Features and Benefits• Maximum frequency: 18 GHz/ 26.5 GHz/ 50 GHz• 1 to 4 SPDT switches or 1 to 2 SP6T switch configurations• SCPI Controllable via VISA and EasySSU software• USB Connectivity• Size: WxHxD=153x62.4x137.5 mm• RF connector: SMA female or 2.4 mm female

Model and Main index
Model

SSU5181A/SSU5182A
SSU5183A/SSU5184A

SSU5261A/SSU5262A
SSU5263A/SSU5264A

SSU5265A
SSU5266A

SSU5501A/SSU5502A
SSU5503A/SSU5504A

Frequency range

DC ~ 18 GHz

DC ~ 26.5 GHz

DC ~ 26.5 GHz

DC ~ 50 GHz

Number of Switches

1/2/3/4

1/2/3/4

1/2

1/2/3/4

Switch Type

SPDT

SPDT

SP6T

SPDT

RF connector

SMA female

Driving voltage

12 V

Maximum driving current

1.25 A

Size

W×H×D = 153×62.4×137.5 mm

Weight885 g

www.siglent.com 14

2.4 mm female

SSU5000A Mechanical Switch
Ordering Information
Model

Product Description

SSU5181A

DC ~ 18 GHz, including one SPDT mechanical switch

SSU5182A

DC ~ 18 GHz, including two SPDT mechanical switchs

SSU5183A

DC ~ 18 GHz, including three SPDT mechanical switchs

SSU5184A

DC ~ 18 GHz, including four SPDT mechanical switchs

SSU5261A

DC ~ 26.5 GHz, including one SPDT mechanical switch

SSU5262A

DC ~ 26.5 GHz, including two SPDT mechanical switchs

SSU5263A

DC ~ 26.5 GHz, including three SPDT mechanical switchs

SSU5264A

DC ~ 26.5 GHz, including four SPDT mechanical switchs

SSU5265A

DC ~ 26.5 GHz, including one SP6T mechanical switch

SSU5266A

DC ~ 26.5 GHz, including two SP6T mechanical switchs

SSU5501A

DC ~ 50 GHz, including one SPDT mechanical switch

SSU5502A

DC ~ 50 GHz, including two SPDT mechanical switchs

SSU5503A

DC ~ 50 GHz, including three SPDT mechanical switchs

SSU5504A

DC ~ 50 GHz, including four SPDT mechanical switchs

Standard Configurations

Quantity

USB-cable

1

Quick-start

1

AC-DC adapter

1Power cord

1

Certificate of qualification1

www.siglent.com 15

SSA5000A

Spectrum Analyzer

Features and Benefits• Spectrum Analyzer Frequency Range from 9 kHz up to 13.6 GHz/26.5 GHz• -165 dBm/Hz Displayed Average Noise Level (Typ.)• -105 dBc/Hz@1 GHz,10 kHz offset SSB Phase Noise (Typ.)• 25 MHz/40 MHz Analysis Bandwidth• 100% POI 7.20 μs, Dynamic Range 60 dB, Multi-view for Density, Spectrogram and PvT• Channel power, ACPR, OBW, Harmonic, TOI measurement etc.
• Analog Modulation Analysis and Vector Digital Modulation Analysis• 12.1 inch Multi-Touch Screen, HDMI output• Web Browser Remote Control on PC and Mobile Terminals and File Operation

Design features• Real Time Analysis Mode

• Modulation Analysis Mode

Multi-view and dimensions to monitor complex signals

AM/FM/PM analog modulation , and ASK/FSK/PSK/MSK/QAM vector
modulation a nalysis

www.siglent.com 16

SSA5000A Spectrum Analyzer• Bluetooth Signal Modulation Analysis• Noise Figure Mode

Provide transmitter tests of Low Energy, Basic Rate and Enhanced Data

The noise figure of systems such as amplifiers and frequency converters

Rate Bluetooth standard• Phase Noise Analysis• Pulse Measurement Analysis

One-sided band phase noise test analysis, jitter analysis, etc

Measure the time and frequency parameters of the RF pulses

Model and Main index
Model

SSA5083A

SSA5085A

Frequency Range

9 kHz~13.6 GHz

9 kHz~26.5 GHz

Displayed Average Noise Level-165 dBm/Hz

SSB Phase Noise<-105 dBc/Hz

Analysis Bandwidth

25 MHz, 40 MHz (opt.)

www.siglent.com 17

SSA5000A Spectrum Analyzer
Ordering Information
Product
Product CodeStandard

Options

Accessories

www.siglent.com 18

Description

Order Number

Spectrum Analyzer, 9 kHz ~ 13.6 GHz

SSA5083A

Spectrum Analyzer, 9 kHz ~ 26.5 GHz

SSA5085A

Quick Start, USB Cable, Power Cord, Wireless Mouse, 2.92F-2.92F-40A
SSA5083A upgrade to SSA5085A

SSA5000-F5

Pre-Amplifier, 9 kHz ~ 13.6 GHz

SSA5000-P3

Pre-Amplifier, 9 kHz ~ 26.5 GHz

SSA5000-P5

40 MHz analysis bandwidth

SSA5000-B40

Real-Time Spectrum Analysis

SSA5000-RTA1

Advanced Measurement Kit

SSA5000-AMK

IQ Data Acquisition

SSA5000-IQA

Pulse Measurement

SSA5000-PU

Phase Noise Measurement

SSA5000-PN

Noise Figure Measurement

SSA5000-NF

Analog Modulation Analysis

SSA5000-AMA

Digital Modulation Analysis

SSA5000-DMA

Bluetooth Analysis

SSA5000-BT

EMI Measurement

SSA5000-EMI

IF Output

SSA5000-IFO

OCXO Precise Reference source, Factory installed

10M_OCXO_L

2.92mm(F)-2.92mm(F) adaptor, DC ~ 40 GHz

2.92F-2.92F-40A

N(M)-N(M) cable, DC ~ 18 GHz, 1000 mm

N-N-18L

N(M)-SMA(M) cable, 18 GHz, 1000 mm

N-SMA-18L

SMA(M)-SMA(M) cable, 18 GHz, 1000 mm

SMA-SMA-18L

SMA(M)-SMA(M) cable, 26.5 GHz, 1000 mm

SMA-SMA-26L

SMA(F)-SMA(M) cable, 26.5 GHz, 1000 mm

SMAF-SMA-26L

USB-GPIB Adaptor

USB-GPIB

300 kHz~3 GHz Near Field Probe Kit:
3 H-probes (20/10/5 mm), 1 E-probe (5 mm)

SRF5030TSSA3000X-R
Real-Time
Spectrum Analyzer

Features and Benefits• Spectrum Analyzer Frequency Range from 9 kHz up to 7.5 GHz• Vector Network Analyzer Frequency Range from 100 kHz up to 7.5 GHz• -165 dBm/Hz Displayed Average Noise Level (Typ.)• -98 dBc/Hz.@10 kHz Offset Phase Noise (1 GHz, Typ.)• Level Measurement Uncertainty < 0.7 dB (Typ.)• 1 Hz Minimum Resolution Bandwidth (RBW)• Preamplifier and Tracking Generator Standard• Up to 40 MHz Real Time Analysis Bandwidth (Opt.)• 100% POI 7.20 μs, Dynamic Range 60 dB, Multi-view for Density, Spectrogram, PvT and 3D• Distance To Fault• Advanced Measurement Kit (Opt.)• Modulation Analysis Mode (Opt.)• EMI Measurement Mode (Opt.)• 10.1 inch Multi-Touch Screen , Mouse and Keyboard supported• Web Browser Remote Control on PC and Mobile Terminals and File Operationwww.siglent.com 19

SSA3000X-R Real-Time Spectrum Analyzer
Design features• 10.1 lnch Display with Multi-Touch Screen• Phase noise <-98 dBc/Hz@1 GHz• Minimum 1 Hz Resolution Bandwidth (RBW)• ACPR in Advanced Measurement Kit• -165 dBm/Hz Displayed Average Noise Level• Monitor in Advanced Measurement Kit

www.siglent.com 20

SSA3000X-R Real-Time Spectrum Analyzer• Modulation Analysis Mode

• Real Time Analysis Mode

AM/FM, ASK/FSK/PSK/MSK/QAM Vector Signal Modulation Analysis andDensity,3D,Spectrogram,PvT,Multi-view and dimensions to monitor

EVM evaluation. The analysis BW is same with real-time BW in RTSA

complex signals

mode• Vector Network Analyzer Mode

• EMI Measurement Mode

100 kHz~7.5 GHz Vector S11 and S21 measurement, Multi Formats

EMI Measurement with CISPR 16-1-1 EMI filter, Quasi-peak Detector, andOverlay Display

pre-stored standards

Model and Main index
Model

SSA3032X-R

SSA3050X-R

SSA3075X-R

Frequency Range

9 kHz~3.2 GHz

9 kHz~5.0 GHz

9 kHz~7.5 GHz

Resolution Bandwidth

1 Hz~3 MHz

1 Hz~3 MHz

1 Hz~3 MHz

Displayed Average Noise Level-165 dBm/Hz-165 dBm/Hz-165 dBm/Hz

SSB Phase Noise<-98 dBc/Hz<-98 dBc/Hz<-98 dBc/Hz

Third-order intercept(TOI)+14 dbm+14 dbm+14 dbm

Total Amplitude Accuracy< 0.7 dB

< 0.7 dB

< 0.7 dB

Tracking Generator

100 kHz~3.2 GHz

100 kHz~5.0 GHz

100 kHz~7.5 GHz

Real Time Band Width

25 MHz, 40 MHz (Option)

RTSA SFDR

60 dB

100% POI

7.20 μs

RTSA Measurement

Density, Spectrogram, 3D, PvT

VNA measurement

Vector S11, Vector S21

VNA Dynamic Range

90 dB

Distance to Fault

Timing Domain Analysis Locator

Touch Screen

Multi Touch, Mouse and Keyboard supported

Advanced Measurement

CHP, ACPR, OBW, CNR, Harmonic, TOI, Monitor

Modulation Analysis

AM, FM, ASK, FSK, MSK, PSK, QAM

EMI Measurement

EMI Filter and Quasi-Peak Detector, Log Scale and Limit Line

Communication Interface

LAN, USB Device, USB Host (USB-GPIB)

Remote Control Capability

SCPI/Labview/IVI based on USB-TMC/VXI-11/Socket/Telnet

Remote Controller

NI-MAX, Web Browser, Easy Spectrum software, File Explorer
www.siglent.com 21

SSA3000X-R Real-Time Spectrum Analyzer
Ordering Information
Product

Product CodeStandard Accessories

Common Options andAccessories

VNA Options

EMI Measurement Options

Modulation Analysis Options

www.siglent.com 22

Description

Order Number

Real Time Spectrum Analyzer, 9 kHz~3.2 GHz, Preamp and TG standard,
VNA standard

SSA3032X-R

Real Time Spectrum Analyzer, 9 kHz~5.0 GHz, Preamp and TG standard,
VNA standard

SSA3050X-R

Real Time Spectrum Analyzer, 9 kHz~7.5 GHz, Preamp and TG standard,
VNA standard

SSA3075X-R

Quick Start, USB Cable, Power Cord
Advanced Measurement Kit

SSA3000XR-AMK

40 MHz Analysis BandWidth

SSA3000XR-RT40

Utility Kit:
N(M)-SMA(M) cable(6 GHz), N(M)-N(M) cable(6 GHz),
N(M)-BNC(F) adaptor x2, N(M)-SMA(F) adaptor x2, 10 dB 1W attenuator

UKitSSA3X

N(M)-BNC(M) cable, 70 cm, 2 GHz

N-BNC-2L

N(M)-SMA(M) cable, 70 cm, 6 GHz

N-SMA-6L

N(M)-N(M) cable, 70 cm, 6 GHz

N-N-6L

N(M)-SMA(M) cable, 100 cm, 18 GHz

N-SMA-18L

N(M)-N(M) cable, 100 cm, 18 GHz

N-N-18L

SMA(M)-SMA(M) cable, 100 cm, 18 GHz

SMA-SMA-18L

USB-GPIB Adaptor

USB-GPIBSoft carrying bag

BAG-S2

6U Rack Mount Kit

SSA-RMKN type Economic Calibration Kit, DC~4.5 GHz, 50 Ω

F503MEN type Economic Calibration Kit, DC~4.5 GHz, 50 Ω

F503FE

3.5 mm type Economic Calibration Kit, DC~4.5 GHz, 50 Ω

F603ME

3.5 mm type Economic Calibration Kit, DC~4.5 GHz, 50 Ω

F603FEN type Standard Calibration Kit, DC~9 GHz, 50 Ω

F504MSN type Standard Calibration Kit, DC~9 GHz, 50 Ω

F504FS

3.5 mm type Standard Calibration Kit, DC~9 GHz, 50 Ω

F604MS

3.5 mm type Standard Calibration Kit, DC~9 GHz, 50 Ω

F604FSEMI Measurement Mode

SSA3000XR-EMI

300 kHz~3 GHz Near Field Probe Kit:
3 H-probes (20/10/5 mm), 1 E-probe (5 mm)

SRF5030T

Analog Modulation Analysis: AM, FM

SSA3000XR-AMA

Digital Modulation Analysis: ASK, FSK, MSK, PSK, QAM

SSA3000XR-WDMASSA3000X Plus
Spectrum Analyzer

Features and Benefits• Frequency Range from 9 kHz up to 7.5 GHz• -165 dBm/Hz Displayed Average Noise Level (Typ.)• -98 dBc/Hz.@10 kHz Offset Phase Noise (1 GHz, Typ.)• Level Measurement Uncertainty < 0.7 dB (Typ.)• 1 Hz Minimum Resolution Bandwidth (RBW)• Preamplifier Standard• Tracking Generator included at no charge• Vector Signal Modulation Analysis (Opt.)• EMI Filter and Quasi-Peak Detector (Opt.)• Advanced Measurement Kit (Opt.)• 10.1 lnch Multi-Touch Screen , Mouse and Keyboard supported• Web Browser Remote Control on PC and Mobile Terminals and File Operation

Design features• 10.1 lnch (1024x600) Touch Screen• Mininmum 1 Hz Resolution Bandwidth (RBW)

www.siglent.com 23

SSA3000X Plus Spectrum Analyzer• -165 dBm/Hz Displayed Average Noise Level• Phase noise <-98 dBc/Hz@1 GHz• Advanced Measurement Kit• Spectrum Monitor in Advanced Measurement
Kit• Modulation Analysis Mode

• EMI Measurement Mode

AM/FM, ASK/FSK/PSK/MSK/QAM Vector Signal Modulation Analysis, EVM

EMI Measurement with CISPR 16-1-1 EMI filter, Quasi-peak Detector, andevaluation

pre-stored standards

www.siglent.com 24

SSA3000X Plus Spectrum Analyzer
Model and Main index
Model

SSA3015X Plus

SSA3021X Plus

SSA3032X Plus

SSA3075X Plus

Frequency Range

9 kHz~1.5 GHz

9 kHz~2.1 GHz

9 kHz~3.2 GHz

9 kHz~7.5 GHz

Resolution Bandwidth

1 Hz~1 MHz

1 Hz~1 MHz

1 Hz~1 MHz

1 Hz~3 MHz

Displayed Average Noise Level-156 dBm/Hz-161 dBm/Hz-161 dBm/Hz-165 dBm/Hz

SSB Phase Noise< -99 dBc/Hz< -98 dBc/Hz<-98 dBc/Hz<-98 dBc/Hz

Third-order intercept+10 dBm+10 dBm+10 dBm+14 dBm

Total Amplitude Accuracy< 1.2 dB

< 0.7 dB

< 0.7 dB

< 0.7 dB

Tracking Generator

100 kHz~1.5 GHz

100 kHz~2.1 GHz

100 kHz~3.2 GHz

100 kHz~7.5 GHz

Touch Screen

Multi Touch, Mouse and Keyboard supported

Advanced Measurement

CHP, ACPR, OBW, CNR, Harmonic, TOI, Monitor

Reflection Measurement

VSWR measurement using Reflection Bridge

EMI Test

EMI Filter and Quasi-Peak Detector, Log Scale and Limit Line

Modulation Analysis

AM, FM; ASK, FSK, MSK, PSK, QAM

Communication Interface

LAN, USB Device, USB Host (USB-GPIB)

Remote Control Capability

SCPI/Labview/IVI based on USB-TMC/VXI-11/Socket/Telnet

Remote Controller

NI-MAX, Web Browser, Easy Spectrum software, File Explorer

Ordering Information
Product

Product CodeStandard Accessories

Common Options and Accessories

Reflection Measurement Options

Description

Order Number

Spectrum Analyzer, 9 kHz ~ 1.5 GHz

SSA3015X Plus

Spectrum Analyzer, 9 kHz ~ 2.1 GHz

SSA3021X Plus

Spectrum Analyzer, 9 kHz ~ 3.2 GHz

SSA3032X Plus

Spectrum Analyzer, 9 kHz ~ 7.5 GHz

SSA3075X Plus

Quick Start, USB Cable, Power Cord
Tracking Generator

SSA3000XP-TG

Advanced Measurement Kit

SSA3000XP-AMK

Utility Kit:
N (M)-SMA (M) cable (6 GHz), N (M)-N (M) cable(6 GHz), N (M)-BNC (F) adaptor x2, N (M)-SMA (F)adaptor x2, 10 dB 1W attenuator

UKitSSA3X

N (M)-BNC (M) cable, 70 cm, 2 GHz

N-BNC-2L

N (M)-SMA (M) cable, 70 cm, 6 GHz

N-SMA-6L

N (M)-N (M) cable, 70 cm, 6 GHz

N-N-6LN (M)-SMA (M) cable, 100 cm, 18 GHz

N-SMA-18L

N (M)-N (M) cable, 100 cm, 18 GHz

N-N-18L

SMA (M)-SMA (M) cable, 100 cm, 18 GHz

SMA-SMA-18L

USB-GPIB Adaptor

USB-GPIBSoft carrying bag

BAG-S2

6U Rack Mount Kit

SSA-RMK

Tracking Generator

SSA3000XP-TG

Reflection Measurement

SSA3000-Refl

Reflection Bridge Kit:
Reflection Bridge (1 MHz~2.5 GHz), N(M)-N(M)adaptors x2

RB3X2550 Ω, N type Male, 4.5 GHz Economic Calibration Kit:
F503ME
Open(M), Short(M), Match(M), Through Adapter(F-F)
EMI test Options

Modulation Analysis Options

EMI Measurement Mode

SSA3000XP-EMI

300 kHz~3 GHz Near Field Probe Kit:
3 H-probes (20/10/5 mm), 1 E-probe (5 mm)

SRF5030T

Digital Modulation: ASK, FSK, MSK, PSK, QAM

SSA3000XP-DMA

Analog Modulation: AM, FM

SSA3000XP-AMA
www.siglent.com 25

SSA3000X

Spectrum Analyzer

Features and Benefits• All-Digital IF Technology• Frequency Range from 9 kHz up to 3.2 GHz• -161 dBm/Hz Displayed Average Noise Level (Typ.)• -98 dBc/Hz @10 kHz Offset Phase Noise (1 GHz, Typ.)• Total Amplitude Accuracy < 0.7 dB• 1 Hz Minimum Resolution Bandwidth (RBW)• Preamplifier Standard• Up to 3.2 GHz Tracking Generator Kit• Reflection Measurement Kit (Opt.)• Advanced Measurement Kit (Opt.)• EMI Pre-compliance Measurements Kit (Opt.)• 10.1 lnch WVGA (1024x600) Display

Design features• Support four traces and cursors independently

www.siglent.com 26• 1 Hz Minimum Resolution Bandwidth (RBW)SSA3000X Spectrum Analyzer• -151 dBm Displayed Average Noise Level• Phase noise -98 dBc/Hz@1 GHz, offset 10 kHz(RBW=10 Hz)• Demodulation at the zero span

• Advanced power measurement, calculate
the ACPR parameters• Reflection measurement, acquirecharacteristic curve of the Return Loss

• EMI filter, Quasi-Peak detector following
CISPR 16

Model and Main index
Model

SSA3021X

SSA3032X

Frequency Range

9 kHz~2.1 GHz

9 kHz~3.2 GHz

Resolution Bandwidth

1 Hz~1 MHz, in 1-3-10 sequence

1 Hz~1 MHz, in 1-3-10 sequence

Displayed Average Noise Level-161 dBm/Hz, Normalize to 1 Hz (typ.)-161 dBm/Hz, Normalize to 1 Hz (typ.)Phase Noise<-98 dBc/Hz@1 GHz, 10 kHz offset<-98 dBc/Hz@1 GHz, 10 kHz offset

Amplitude Precision< 0.7 dB

< 0.7 dB
www.siglent.com 27

SSA3000X Spectrum Analyzer
Ordering Information
Product Description
Product code
Standard configurations

Utility Options

EMI Options

Reflect Measurement Options

www.siglent.com 28

SSA3000X Spectrum Analyzer

Order Number

Spectrum Analyzer, 9 kHz~3.2 GHz

SSA3032X

Spectrum Analyzer, 9 kHz~2.1 GHz

SSA3021X

A Quick Start, A USB Cable, A Power Cord, A Calibration
QG-SSA3000X
Certificate
Tracking Generator Kit

TG-SSA3000X

Advanced Measurement Kit

AMK-SSA3000X

Utility Kit:
N(M)-SMA(M) cable
N(M)-N(M) cable
N(M)-BNC(F) adaptor(2 pcs)
N(M)-SMA(F) adaptor(2 pcs)
10 dB attenuator

UKitSSA3X

N(M)-SMA(M) cable

N-SMA-6L

N(M)-N(M) cable

N-N-6L

N(M)-BNC(M) cable

N-BNC-2L

Soft carrying bag

BAG-SCC

Rack Mount Kit

SSA-RMKEMI Measurement Kit:
EMI Filter and Quasi Peak Detector,
EMI test option in EasySpectrum Software

EMI-SSA3000X

Near Field Probe:H field probe sets(25 mm, 10 mm, 5 mm,2mm), 30 MHz~3.0 GHz

SRF5030

Near Field Probe:H field probe sets(20 mm, 10 mm, 5 mm) ,E field probe (5 mm), 300 kHz~3.0 GHz

SRF5030T

Tracking Generator Kit

TG-SSA3000X

Reflect Measurement Kit

Refl-SSA3000X

VSWR Bridge Kit:
including Refl-SSA3000X
VSWR Bridge(1 MHz~2 GHz)
N(M)-N(M) adaptor(2 pcs)

RBSSA3X20

SHA850A

Portable Spectrum
Analyzer

Features and Benefits• Spectrum Analyzer Frequency Range from 9 kHz up to 7.5 GHz, -165 dBm/Hz Displayed Average Noise Level (Typ.), -104 dBc/Hz.@10 kHz
Offset Phase Noise (1 GHz, Typ.), 1 Hz up to 10 MHz Minimum Resolution Bandwidth (RBW), Preamplifier and independent signal source up
to 7.5 GHz, GPS positioning and logging• Cable and Antenna Test Frequency Range from 100 kHz up to 7.5 GHz, Distance To Fault and Time Domain Analysis• Vector Network Analyzer, Bias out up to 32VDC• Typical working time 4 hours, 3.2 kg net weight, 8.4 lnch Multi-Touch Screen , Mouse and Keyboard supported

Design features• Spectrum Analyzer 8.4 inch multi-touch screen
and full keyboard control• Channel Power and ACPR measurement

www.siglent.com 29

SHA850A Handheld Spectrum Analyzer• GPS Location and trace log recorder, sync 10MHz reference clock• Interference analysis with directional antenna• Vector Network Analyzer• Modulation Analysis

100 kHz-7.5 GHz Vector S11 and S21 measurement, Multi Formats

AM/FM/PM analog modulation, and ASK/FSK/PSK/MSK/QAM digital

Overlay Display

modulation analysis• Cable and Antenna Test
Cable and Antenna Test based on Timing Domain Analysis

www.siglent.com 30

SHA850A Handheld Spectrum Analyzer
Model and Main index
Model

SHA851A

SHA852A

Spectrum Analyzer

9 kHz~3.6 GHz

9 kHz~7.5 GHz

Cable and Antenna Test

100 kHz~3.6 GHz

100 kHz~7.5 GHz

Ordering Information
Product
Product code
Standard Accessories

Options

General Accessories

CAT&VNA Accessories

VNA Options

Description

Order Number

Spectrum & Vector Network Analyzer, 9 kHz~3.6 GHz

SHA851A

Spectrum & Vector Network Analyzer, 9 kHz~7.5 GHz

SHA852A

Quick Start, USB type-C cable, Power cord, AC-DC adapter, Rechargeable lithium battery, Portable bag
SHA851A to SHA852A

SHA850-F2

Source

SHA850-SOR

Vector Network Analysis

SHA850-VNA

Advanced Measurement Kit

SHA850-AMK

Analog Modulation Analysis

SHA850-AMA

Digital Modulation Analysis

SHA850-DMA

DC Bias Out

SHA850-BIAS

GPS Receiver

SHA850-GPS

GPS Logging(need GPS Receiver)

SHA850-GPSM

Rechargeable lithium battery

10V8_BAT

AC-DC adapter

12V_AP_4A

Portable bag

BAG-H2GPS antenna, SMA(M), 100 cm

ANT-GPS1

S5000 Directional Antenna Suit:
S5001-VHF (10 MHz~200 MHz), S5001-UHF(200 MHz~500 MHz), S5001-LP (500 MHz~8 GHz),
Preamp (10 dB, 9 kHz~8 GHz)

ANT-DA1

Near field probe kit: 300 kHz~3 GHz,
H-field probes (20 mm,10 mm,5 mm), E-field probe (5 mm)

SRF5030T

Utility Kit:
N(M)-SMA(M) cable(6 GHz), N(M)-N(M) cable(6 GHz),
N(M)-BNC(F) adaptor x2, N(M)-SMA(F) adaptor x2,
10 dB 1W attenuator

UKitSSA3X

N(M)-BNC(M) cable, DC~2 GHz, 700 mm

N-BNC-2L

N(M)-SMA(M) cable, DC~6 GHz, 700 mm

N-SMA-6L

N(M)-N(M) cable, DC~6 GHz, 700 mm

N-N-6L

N(M)-N(M) cable ,DC~18 GHz, 1000 mm

N-N-18L

N(M)-SMA(M) cable ,DC~18 GHz, 1000 mm

N-SMA-18L

SMA(M)-SMA(M) cable ,DC~18 GHz, 1000 mm

SMA-SMA-18L

N type Integrated Calibration Kit, Male, DC~9 GHz,50 Ω

Y504MSN type Integrated Calibration Kit, Female, DC~9 GHz,50 Ω

Y504FSN type Precision Calibration Kit, DC~9 GHz, 50 Ω

F504TS

3.5 mm type Precision Calibration Kit, DC~9 GHz, 50 Ω

F604TS

2 ports, 9 kHz ~ 4.5 GHz, SMA female

SEM5002A

2 ports, 9 kHz ~ 9 GHz, SMA female

SEM5012A

2 ports, 100 kHz ~ 13.5 GHz, 3.5 mm female

SEM5022A

2 ports, 100 kHz ~ 26.5 GHz, 3.5 mm female

SEM5032A

4 ports, 9 kHz ~ 4.5 GHz, SMA female

SEM5004A

4 ports, 9 kHz ~ 9 GHz, SMA female

SEM5014A

4 ports, 100 kHz ~ 13.5 GHz, 3.5 mm female

SEM5024A

4 ports, 100 kHz ~ 26.5 GHz, 3.5 mm female

SEM5034A
www.siglent.com 31

SVA1000X

Spectrum & Vector
Network Analyzer

Features and Benefits• Spectrum Analyzer Frequency Range from 9 kHz up to 7.5 GHz• Vector Network Analyzer Frequency Range from 100 kHz up to 7.5 GHz• -165 dBm/Hz Displayed Average Noise Level (Typ.)• -98 dBc/Hz.@10 kHz Offset Phase Noise (1 GHz, Typ.)• Level Measurement Uncertainty < 0.7 dB (Typ.)• 1 Hz Minimum Resolution Bandwidth (RBW)• Preamplifier Standard• Tracking Generator Standard• Distance To Fault (Opt.)• Vector Signal Modulation Analysis (Opt.)• EMI Filter and Quasi-Peak Detector Kit (Opt.)• Advanced Measurement Kit (Opt.)• 10.1 lnch Multi-Touch Screen , Mouse and Keyboard supported• Web Browser Remote Control on PC and Mobile Terminals and File Operationwww.siglent.com 32

SVA1000X Spectrum & Vector Network Analyzer
Design features• 10.1 lnch Display with Multi-Touch Screen• Phase noise <-98 dBc/Hz@1 GHz, offset 10 kHz• Minimum 1 Hz Resolution Bandwidth (RBW)• ACPR in Advanced Measurement Kit• -165 dBm/Hz Displayed Average Noise Level• Monitor in Advanced Measurement Kit• 100 k-7.5 GHz Vector S11 and S21
measurement, Multi Formats Overlay Display• Cable and Antenna Test based on Timing
Domain Analysis

www.siglent.com 33

SVA1000X Spectrum & Vector Network Analyzer• Modulation Analysis Mode

• EMI Measurement Mode

AM/FM, ASK/FSK/PSK/MSK/QAM Vector Signal Modulation Analysis, EVM

EMI Measurement with CISPR 16-1-1 EMI filter, Quasi-peak Detector, andevaluation

pre-stored standards

Model and Main index
Model

SVA1015X

SVA1032X

SVA1075X

Spectrum Analyzer Frequency Range

9 kHz~1.5 GHz

9 kHz~3.2 GHz

9 kHz~7.5 GHz

Vector Network Analyzer Frequency
Range

100 kHz~1.5 GHz

100 kHz~3.2 GHz

100 kHz~7.5 GHz

Resolution Bandwidth

1 Hz~1 MHz

1 Hz~1 MHz

1 Hz~3 MHz

Displayed Average Noise Level-156 dBm/Hz-161 dBm/Hz-165 dBm/Hz

SSB Phase Noise<-99 dBc/Hz<-98 dBc/Hz<-98 dBc/Hz

Total Amplitude Accuracy< 1.2 dB

< 0.7 dB

< 0.7 dB

Tracking Generator

100 kHz~1.5 GHz

100 kHz~3.2 GHz

100 kHz~7.5 GHz

VNA measurement

Vector S11,Vector S21

Distance to Fault

VNA Timing Domain Analysis Locator

Touch Screen

Multi Touch, Mouse and Keyboard supported

Advanced Measurement

CHP, ACPR, OBW, CNR, Harmonic, TOI, Monitor

Reflection Measurement

VSWR measurement using Reflection Bridge

EMI Test

EMI Filter and Quasi-Peak Detector, Log Scale and Limit Line

Modulation Analysis

AM, FM; ASK, FSK, MSK, PSK, QAM

Communication Interface

LAN, USB Device, USB Host (USB-GPIB)

Remote Control Capability

SCPI/Labview/IVI based on USB-TMC/VXI-11/Socket/Telnet

Remote Controller

NI-MAX, Web Browser, Easy Spectrum software, File Explorer

www.siglent.com 34

SVA1000X Spectrum & Vector Network Analyzer
Ordering Information
Product
Product CodeStandard Accessories

Common Options andAccessories

VNA Options

EMI test Options

Modulation Analysis Options

VNA Options

Description

Order Number

Spectrum & Vector Network Analyzer, 1.5 GHz

SVA1015X

Spectrum & Vector Network Analyzer, 3.2 GHz

SVA1032X

Spectrum & Vector Network Analyzer, 7.5 GHz

SVA1075X

Quick Start, USB Cable, Power Cord
Advanced Measurement Kit

SVA1000X-AMK

Utility Kit:
N (M)-SMA (M) cable (6 GHz), N (M)-N (M) cable (6 GHz),
N (M)-BNC (F) adaptor x 2, N (M)-SMA (F) adaptor x 2,
10 dB 1W attenuator

UKitSSA3X

N (M)-SMA (M) cable, 70 cm, 6 GHz

N-SMA-6L

N (M)-N (M) cable, 70 cm, 6 GHz

N-N-6LN (M)-BNC (M) cable, 70 cm, 2 GHz

N-BNC-2L

N (M)-N (M) cable, 100 cm, 18 GHz

N-N-18L

N (M)-SMA (M) cable, 100 cm, 18 GHz

N-SMA-18L

SMA(M)-SMA(M) cable, 100 cm, 18 GHz

SMA-SMA-18L

USB-GPIB Adaptor

USB-GPIBSoft carrying bag

BAG-S2

6U Rack Mount Kit

SSA-RMKDistance To Fault

SVA1000X-DTF

Mechanical Calibration Kit: Open (M), Short (M), Match (M,50),
Through (F-F), 4.5 GHz, N-Male connector

F503ME

Mechanical Calibration Kit: OSLT, DC - 4.5 GHz, N-Female
connector

F503FE

Mechanical Calibration Kit: OSLT, DC - 4.5 GHz, 3.5mm
SMA-Male connector

F603ME

Mechanical Calibration Kit: Open (M), Short (M), Match (M,50),
Through (F-F), 4.5 GHz, SMA-Female connector

F603FE

Mechanical Calibration Kit: OSLT, DC - 9 GHz, N-Maleconnector

F504MS

Mechanical Calibration Kit: OSLT, DC - 9 GHz, N-Female
connector

F504FS

Mechanical Calibration Kit: OSLT, DC - 9 GHz, 3.5 mm
SMA-Male connector

F604MS

Mechanical Calibration Kit: OSLT, DC - 9 GHz, 3.5 mm
SMA-Female connector

F604FS

N-type, Male and Female, 50 Ω Calibration Kit, 0~9 GHz

F504TS

3.5 mm, Male and Female, 50 Ω Calibration Kit, 0~9 GHz

F604TSEMI Measurement Kit:
EMI Filter and Quasi Peak Detector,
EMI Receiver Mode in EasySpectrum Software

SVA1000X-EMI

300 kHz~3 GHz Near Field Probe Kit:
3 H-probes (20/10/5 mm), 1 E-probe (5 mm)

SRF5030T

Digital Modulation: ASK, FSK, MSK, PSK, QAM

SVA1000X-DMA

Analog Modulation: AM, FM

SVA1000X-AMA

2 ports, 9 kHz ~ 4.5 GHz, SMA female

SEM5002A

2 ports, 9 kHz ~ 9 GHz, SMA female

SEM5012A

2 ports, 100 kHz ~ 13.5 GHz, 3.5 mm female

SEM5022A

2 ports, 100 kHz ~ 26.5 GHz, 3.5 mm female

SEM5032A

4 ports, 9 kHz ~ 4.5 GHz, SMA female

SEM5004A

4 ports, 9 kHz ~ 9 GHz, SMA female

SEM5014A

4 ports, 100 kHz ~ 13.5 GHz, 3.5 mm female

SEM5024A

4 ports, 100 kHz ~ 26.5 GHz, 3.5 mm female

SEM5034A
www.siglent.com 35

SSG6000A

RF Signal Generator

Features and Benefits• Frequency up to 13.6 GHz/ 20 GHz/ 40GHz• 0.01 Hz frequency setting resolution• Level setting range: -130 dBm ~ 24 dBm• Phase Noise: -135 dBc / Hz @ 1 GHz, 20 kHz offset (typ.)• Level error ≤ 0.7 dB (typ.)• Provides AM analog modulation with internal, external or Int+Ext source• Single pulse, double pulse and pulse train generator (option)• The power meter control kit can easily use the power meter to measure power, control power output and correct line loss
• 5 inch TFT capacitive touch screen, mouse and keyboard supported• Web browser remote control on PC and mobile terminals• Standard interface includes USB Host, USB Device (USB TMC), LAN (VXI-11, Socket, Telnet). Optional interface: GPIB

Design features• LO in Up/Down Converter Measurement

www.siglent.com 36• ADC Measurement

SSG6000A RF Signal Generator• DSO Measurement• DAC Measurement• Receiver Blocking Test

Model and Main index
Model

SSG6083A

SSG6085A

SSG6087A

Frequency Range

CW MODE
100 kHz~13.6 GHz

CW MODE
100 kHz~20 GHz

CW MODE
100 kHz~40 GHz

Frequency Resolution

0.01 Hz

Amplitude Resolution

0.01 dB

Level error≤ 0.7 dB(typ.)Phase noise-135 dBc/Hz @1 GHz, offset 20 kHz (typ.)Display

5 inch capacitance touch screen, RGB (800*480)

www.siglent.com 37

SSG6000A RF Signal Generator

Ordering Information
Product Description
Product code
Standard configurations

optionwww.siglent.com 38

SSG6000A Signal Generator

Order Number

Analog Signal Generator 100 kHz~13.6 GHz

SSG6083A

Analog Signal Generator 100 kHz~20 GHz

SSG6085A

Analog Signal Generator 100 kHz~40 GHz

SSG6087A

Quick start, an USB cable, calibration certificate, power cord, 2.92 mm female to female adapter
Pulse modulation

SSG6080A-PU

Pulse train generator

SSG6080A-PT

Rack mount kit

SSG6000A-RMK

USB-GPIB adapter

USB-GPIB

Upgrade 13.6 GHz to 20 GHz

SSG6080A-F85

SSG5000A

RF Signal Generator

Features and Benefits• Frequency up to 13.6 GHz / 20 GHz• 0.001 Hz frequency setting resolution• Level setting range: -130 dBm ~ 25 dBm• Phase Noise: -120 dBc / Hz @ 1 GHz, 20 kHz offset (typ.)• Level error ≤ 0.7 dB (typ.)• Provides AM, FM, PM analog modulation with internal, external or Int+Ext source• Single pulse, double pulse and pulse train generator (option)• The power meter control kit can easily use the power meter to measure power, control power output and correct line loss
• 5 inch TFT capacitive touch screen, mouse and keyboard supported• Web browser remote control on PC and mobile terminals• Standard interface includes USB Host, USB Device (USB TMC), LAN (VXI-11, Socket, Telnet). Optional interface: GPIB

Model and Main index
Model

SSG5083A

SSG5085A

Frequency Range

CW MODE 9 kHz~13.6 GHz

CW MODE 9 kHz~20 GHz

Frequency Resolution

0.001 Hz

Amplitude Resolution

0.01 dB

Level error≤ 0.7 dB(typ.)Phase noise-120 dBc/Hz @1 GHz, offset 20 kHz (typ.)Display

5 inch capacitance touch screen, RGB (800*480)
www.siglent.com 39

SSG5000A RF Signal Generator
Ordering Information
Product Description
Product code
Standard configurations

option

SSG5000A Signal Generator

Order Number

Analog Signal Generator 9 kHz~13.6 GHz

SSG5083A

Analog Signal Generator 9 kHz~20 GHz

SSG5085A

Quick start, an USB cable, calibration certificate, power cord
Pulse modulation

SSG5080A-PU

Pulse train generator

SSG5080A-PT

110 dB Attenuator module[1]SSG5080A-LP

Rack mount kit

SSG-RMK

USB-GPIB adapter

USB-GPIB

Upgrade 13.6 GHz to 20 GHz

SSG5080A-F85[1] Assembled and calibrated in factory only

www.siglent.com 40

SSG5000X

RF Signal Generator

Features and Benefits• Frequency up to 4 GHz/6 GHz• 0.001 Hz frequency setting resolution• High output power up to +26 dBm (typ.)• Phase Noise: -120 dBc/ Hz @ 1 GHz, 20 kHz offset (typ.)• User flatness correction with power sensor to correct the cable loss
• Provides AM, FM, PM analog modulation with internal, external or Int+Ext source• Single pulse, double pulse and Pulse train generator (option)• Internal IQ modulation with 150 MHz modulation bandwidth with perfect in-factory calibration• Internal include some digital communication stand file such as 5G-NR, LTE, WCDMA, WLAN, and playback them
• Internal Custom mode generate common IQ signal such as QAM, FSK, ASK, MSK• Analog differential I/Q outputs• External analog I/Q input• USB-power meter measurement• 5 inch TFT capacitive touch screen, mouse and keyboard supported• Web browser remote control on PC and mobile terminals• Standard interface included USB Host, USB Device (USB TMC), LAN (VXI-11, Socket, Telnet). Optional interface: GPIB

www.siglent.com 41

SSG5000X RF Signal Generator
Design features• Double pulse modulation• Pulse train generator• Works with Power sensor to use the measured values to compensate the cable losses with internal
control functions• Custom mod can generate IQ modulated signal such as QAM, PSK, ASK, FSK, the maximum sample
rate is 120 Msps/s

www.siglent.com 42

SSG5000X RF Signal Generator• Multi-tone mode to output multi-signal• ARB mode to build and replay waveformsequence• Arb mode to replay back communication stand files

3 GPP WCDMA TM 1-64 DPCH ACPR

• ARB mod to generate multi-carrier

3 GPP WCDMA TM 1-64 DPCH EVM• ARB mod to use the marker to label symbol of
the waveform files and simultaneously output
a pulse from the invent interface, this cansynchronize another device.

www.siglent.com 43

SSG5000X RF Signal Generator• ARB mode to add real time AWGN to the digital IQ to satisfy the receiver performance tests of receiver• ARB mod to clip the signal of the peak power and display the CCDF (cytotoxic cell differentiation factor)• SigIQPro Signal Generation Software (Optional)
SigIQPro is a flexible PC-based signal generation software that takes signal
generation to a whole new level, making it easy to generate complex signalsthat are fully compliant with Bluetooth, IoT and other communication standards.
SIGLENT instruments and SigIQPro signal generation software integrate
simulation, design and test to easily meet the needs of users at all stages ofdesign, R&D, and productionwww.siglent.com 44

SSG5000X RF Signal Generator
Model and Main index
Model
Frequency Range

SSG5040X

SSG5060X

SSG5040X-V

SSG5060X-V

CW MODE 9 kHz~4 GHz

CW MODE 9 kHz~6 GHz

CW MODE 9 kHz~4 GHz

CW MODE 9 kHz~6 GHz

IQ MODE 10 MHz~4 GHz

IQ MODE 10 MHz~6 GHz

Frequency Resolution

0.001 Hz

Amplitude Resolution

0.01 dB

Phase noise-120 dBc/Hz @1 GHz, offset 20 kHz (typ.)Display

5 inch capacitance touch screen, RGB (800*480)

Ordering Information
Product Description

Product code

Standard configurations

option

SSG5000X Signal Generator

Order Number

Analog Signal Generator 9 kHz ~ 4 GHz

SSG5040X

Analog Signal Generator 9 kHz ~ 6 GHz

SSG5060X

Vector Signal Generator 10 MHz ~ 4 GHz

SSG5040X-V

Vector Signal Generator 10 MHz ~ 6 GHz

SSG5060X-V

Quick start, an USB cable, calibration certificate, power cord
Pulse train generator

SSG5000X-PT

Rack mount kit

SSG-RMK

USB-GPIB adapter

USB-GPIB

Upgrade 4 GHz to 6 GHz

SSG5000X_F60

Upgrade IQ bandwidth from 75 MHz to 150 MHz

SSG5000XV_B150

Precision Frequency Reference

10M_OCXO_L [1]Generate IOT waveform at device
SigIQPro for Bluetooth waveform playback license

SSG5000XV-IOT[2]

SigIQPro-BT

SigIQPro for IOT waveform playback license

SigIQPro-IOT

SigIQPro for OFDM waveform playback license

SigIQPro-OFDM[1] Assembled and calibrated in factory only
[2] See the SigIQPro User Manual for details

www.siglent.com 45

SSG3000X

RF Signal Generator

Features and Benefits• Frequency up to 2.1 GHz/3.2 GHz• 0.01 Hz frequency setting resolution• Level output from -110 dBm to +13 dBm• Maximum level up to +20 dBm (typ.)• Phase Noise: -110 dBc/ Hz @ 1 GHz , 20 kHz offset (typ.)• Level accuracy ≤0.7 dB (typ.)• Provides AM, FM & PM analog modulation with internal, external or Int+Ext source• Pulse modulation, on/off ratio ≥70 dBc• Pulse train generator (option)• External IQ modulation with SDG6000X as the baseband IQ signal• USB-power meter measurement• 5 inch TFT capacitive touch screen, mouse and keyboard supported• Web browser remote control on PC and mobile terminals• Standard interface include USB Host, USB Device (USB TMC), LAN (VXI-11, Socket, Telnet). Optional interface: GPIB

Design features• Double pulse modulationwww.siglent.com 46• Pulse train generator

SSG3000X RF Generator• Provides double-tone signal with IQ modulation,• Maximum output level up to +20 dBm

easily do TOI testing• Power output display using USB power sensor• Power output control using USB power sensor• Example for auto level control

www.siglent.com 47

SSG3000X RF Generator• External IQ modulation using the SDG6000X as the baseband source• 5 inch touch screen, keyboard and mouse support

Specifications
Specifications are valid under the following condition: The instrument is within the calibration period, has been stored between 0 and 50℃ for at least 2 hours prior
to use, and has been powered on and warmed up for at least 40 minutes. The specifications include the measurement uncertainty, unless otherwise noted.
Specifications: All products are guaranteed to meet published specifications when operating temperatures from 5 to 45℃ , unless otherwise noted.
Typical(typ.): Performance deemed typical implies that 80 percent of the measurement results will meet the typical published performance with a 95th percentile
confidence level at room temperature (approximately 25℃ ). Typical performance is not warranted and does not include measurement uncertainty.
Nominal(nom.): This value indicate the expected mean or average performance, or an attribute whose performance is by design, such as the 50 ohm connector.

www.siglent.com 48

SSG3000X RF Generator
Model and Main index
Model
Frequency Range

SSG3021X

SSG3032X

SSG3021X-IQE

SSG3032X-IQE

CW MODE 9 kHz~2.1 GHz

CW MODE 9 kHz~3.2 GHz

CW MODE 9 kHz~2.1 GHz

CW MODE 9 kHz~3.2 GHz

IQ MODE 10 MHz~2.1 GHz

IQ MODE 10 MHz~3.2 GHz

Frequency Resolution

0.01 Hz

Amplitude Resolution

0.01 dB

Level accuracy

0.7 dB (typ.)Phase noise-110 dBc/Hz @1 GHz ,offset 20 kHz (typ.)Display

5 inch capacitance touch screen, RGB (800*480)

Ordering Information
Product Description

SSG3000X Signal Generator
Signal Generator 9 kHz~2.1 GHz

Product code
Signal Generator 9 kHz~3.2 GHzStandard configurations

optionOrder Number
SSG3021X
SSG3021X-IQE
SSG3032X
SSG3032X-IQE

quick start, an USB cable, calibration certificate, power cord
pulse train generator

SSG3000X-PT

rack mount kit

SSG-RMK

USB-GPIB adapter

USB-GPIB

Upgrade 2.1 GHz to 3.2 GHz

SSG3000X-21BW32

Upgrade 2.1 GHz to 3.2 GHz (with external IQ)

SSG3000X-IQE-21BW32

www.siglent.com 49

Accessories
Type

Model

Near-field
Probe

SRF5030T

GPIB

USB-GPIB Adapter

PictureSpecifications
Near Field Probe: H field probe sets (20 mm, 10 mm, 5 mm) , E field probe (5 mm),
300 kHz~3.0 GHz; distinguished within 10 cm range of the magnetic field; for EMI
radiation interference and the intensity detector

The USB Device interface extends into the GPIB interface, USB-GPIB adapter can moreeasily complete the task of the operation command through the GPIB, USB follow theUSB2.0 specification, GPIB follow the IEEE488.2 standard

N-BNC-2L

N-N-6L

N-SMA-6L

N-N-18L

N-SMA-18L

Cable

www.siglent.com 50

N-BNC cable for SSA3000X Series; 2 GHz bandwidth

N-N cable for SSA3000X, SSA3000X Plus, SSA3000X-R, SVA1000X Series; 6 GHz
bandwidth

N-SMA cable for SSA3000X, SSA3000X Plus, SSA3000X-R, SVA1000X Series; 6 GHz
bandwidthN(M)-N(M) cable for SSA3000X, SSA3000X Plus, SSA3000X-R, SVA1000X series, 100
cm, 18 GHz bandwidthN(M)-SMA(M) cable for SSA3000X, SSA3000X Plus, SSA3000X-R, SVA1000X series,
100 cm, 18 GHz bandwidth

SMA-SMA-18L

SMA(M)-SMA(M) cable, 18 GHz

SMA-SMA-26L

SMA(M)-SMA(M) cable, 26 GHz

SMAF-SMA-26L

SMA(F)-SMA(M) cable, 26 GHz

2.92F-2.92F-40A

2.92 mm Female - 2.92 mm Female adaptor, 40 GHz

V26-N35MN35F-25IN

NMD 3.5 mm(M) – NMD 3.5 mm(F), 26.5 GHz

V26-N35FA35F-25IN

NMD 3.5 mm(F) – APC 3.5 mm(F), 26.5 GHz

Type

Model

Reflection Bridge

RB3X25

Utility Kit

UKitSSA3X

PictureSpecifications
VSWR bridge: (1 MHz~2.5 GHz), N (M) -N (M) adaptor (2 pcs)

Utility Kit for SSA3000X Series: N (M) -SMA (M) cable, N (M) -N (M)cable, N (M) -BNC (F) adaptor (2 pcs), N (M) -SMA (F) adaptor (2 pcs),
10 dB attenuator

F503ME

Through (F-F), 4.5 GHz

F503FE

Mechanical Calibration Kit: OSLT, DC - 4.5 GHz, N-Female connector

F504MS

Mechanical Calibration Kit: OSLT, DC - 9 GHz, N-Male connector

F504FS

Mechanical Calibration Kit: OSLT, DC - 9 GHz, N-Female connector

F504TS

N-type, Male and Female, 50 Ω Calibration Kit, 0~9 GHz

F505TS

VNA Calibration
Kit

Mechanical Calibration Kit: Open (M), Short (M), Match (M,50),

F603ME

F603FE

F604MS

F604FS

F604TS

F604TY

Mechanical Calibration Kit: OSLT, DC - 18 GHz, N-Male and Female
connector

Mechanical Calibration Kit: OSLT, DC - 4.5 GHz, 3.5 mm SMA-Maleconnector

Mechanical Calibration Kit: Open (M), Short (M), Match (M,50),
Through (F-F), 4.5 GHz, SMA-typeMechanical Calibration Kit: OSLT, DC - 9 GHz, 3.5 mm SMA-Maleconnector

Mechanical Calibration Kit: OSLT, DC - 9 GHz, 3.5 mm SMA-Female
connector

3.5 mm, Male and Female, 50 Ω Calibration Kit, 0~9 GHz

Mechanical Calibration Kit: OSLT, DC - 27 GHz, 3.5 mm-Male and Female
connector

Y504MS

Integrated Mechanical Calibration Kit: OSLT, DC - 9 GHz, N-Male

Y504FS

Integrated Mechanical Calibration Kit: OSLT, DC - 9 GHz, N-Female

www.siglent.com 51

Type

Model

SSA-RMK

Rack Mount

SSG-RMK

SSG6000A-RMK

PictureSpecifications
Rackmount kit , compatible with the SSA3000X,SSA3000X Plus,
SVA1000X,SSA3000X-R model; Height 6U

Rack Mount kit; SSG3000X, SSG5000X, SSG5000A, SDG7000A; Height
3U

Rack Mount kit; SSG6000A; Height 2U

Board integrated with RF components like amplier, mixer, filter for vector

RF Test board

SNA-TB01

Rechargeable
lithium battery

10V8_BAT

10.8V, 74 Wh

ANT-GPS1

GPS antenna, SMA(M), 100 cm

network analyzer demonstration

Directional Antenna Suit, N type,
ANT-DA11 antenna (10 MHz~200 MHz),ANT-DA1

ANT-DA12 antenna (200 MHz~500 MHz),ANT-DA13 antenna (500 MHz~8 GHz),
Amplifier handle 12dB@1GHz(typ.)

Antenna

Contains amplifier handle and 10 MHz ~ 200 MHz antenna. Antenna
ANT-DA11

gain 10 dB (typical value); SWR <1:1.9 (typical value); 50 Ω/N type,female; polarization direction horizontal and verticalContains amplifier handle and 200 MHz ~ 500 MHz antenna. Antenna
ANT-DA12

gain 10 dB (typical value); SWR <1:1.9 (typical value); 50 Ω/N type,female; polarization direction horizontal and verticalContains amplifier handle and 500 MHz ~ 8 GHz antenna. Antenna gain
ANT-DA13

10 dB (typical value); SWR <1:1.9 (typical value); 50 Ω/N type, female;polarization direction horizontal and vertical

ADP-18Adjustable differential TDR probe DC~18 GHz

ADP-26

Adjustable differential TDR probe DC~26.5 GHz

ASP-18Adjustable single-end TDR probe DC~18 GHz

ASP-26

Adjustable single-end TDR probe DC~26.5 GHz

TDR Probe

www.siglent.com 52

Type

Model

AC-DC adapter

12V_AP_4A

BAG-S2

PictureSpecifications

12V, 4A

Soft Carry Case for SDS2000X, SDS5000X, SSA3000X, SVA1000X,
SSA3000X Plus

Carry BagBAG-H2

Soft Carry Case for SHA850A, SHN900A

SEM5002A

2 ports, 9 kHz ~ 4.5 GHz, SMA female

SEM5012A

2 ports, 9 kHz ~ 9 GHz, SMA female

SEM5022A

2 ports, 100 kHz ~ 13.5 GHz, 3.5 mm female

SEM5032A

2 ports, 100 kHz ~ 26.5 GHz, 3.5 mm female

SEM5004A

4 ports, 9 kHz ~ 4.5 GHz, SMA female

SEM5014A

4 ports, 9 kHz ~ 9 GHz, SMA female

SEM5024A

4 ports, 100 kHz ~ 13.5 GHz, 3.5 mm female

SEM5034A

4 ports, 100 kHz ~ 26.5 GHz, 3.5 mm female

SSM5122A

2 input ports, 12 output ports, 3.5 mm female, 9 kHz ~ 9 GHz

SSM5124A

2 input ports, 24 output ports, 3.5 mm female, 9 kHz ~ 9 GHz

SSM5142A

4 input ports, 12 output ports, 3.5 mm female, 9 kHz ~ 9 GHz

SSM5144A

4 input ports, 24 output ports, 3.5 mm female, 9 kHz ~ 9 GHz

SSM5321A

2 input ports, 6 output ports, 3.5 mm female, 100 kHz ~ 26.5 GHz

SSM5342A

4 input ports, 12 output ports, 3.5 mm female, 100 kHz ~ 26.5 GHz

VNA Calibration
Kit

Switch Matrix

www.siglent.com 53

Type

Mechanical
SwitchModelSpecifications

SSU5181A

DC ~ 18 GHz, including one SPDT mechanical switch, SMA female

SSU5182A

DC ~ 18 GHz, including two SPDT mechanical switchs, SMA female

SSU5183A

DC ~ 18 GHz, including three SPDT mechanical switchs, SMA female

SSU5184A

DC ~ 18 GHz, including four SPDT mechanical switchs, SMA female

SSU5261A

DC ~ 26.5 GHz, including one SPDT mechanical switch, SMA female

SSU5262A

DC ~ 26.5 GHz, including two SPDT mechanical switchs, SMA female

SSU5263A

DC ~ 26.5 GHz, including three SPDT mechanical switchs, SMA female

SSU5264A

DC ~ 26.5 GHz, including four SPDT mechanical switchs, SMA female

SSU5265A

DC ~ 26.5 GHz, including one SP6T mechanical switch, SMA female

SSU5266A

DC ~ 26.5 GHz, including two SP6T mechanical switchs, SMA female

SSU5501A

DC ~ 50 GHz, including one SPDT mechanical switch, 2.4 mm female

SSU5502A

DC ~ 50 GHz, including two SPDT mechanical switchs, 2.4 mm female

SSU5503A

DC ~ 50 GHz, including three SPDT mechanical switchs, 2.4 mm female

SSU5504A

DC ~ 50 GHz, including four SPDT mechanical switchs, 2.4 mm female

PC Software

SigIQPro

Noise Source
Driver

NSD28

www.siglent.com 54

PictureA comprehensive PC-based software for general and standards-based
signals creation, supporting Bluetooth, IoT, Custom OFDM, etc

Noise source driver, connect spectrum analyzer to noise sourceOther Products Overview
SIGLENT also provides other instruments like Oscilloscopes, AWG, Multimeters, Electronic loads, power supplies.

※ Oscilloscopes ※SDS2000X
plus

SDS7000A

SDS6000A

SDS6000L

SDS5000X

SDS3000X HD SDS2000X HD

Bandwidth

3 GHz ~ 4 GHz

350 MHz ~ 2 GHz

500 MHz ~ 2 GHz

350 MHz ~ 1 GHz

350 MHz ~ 1 GHz

100 MHz ~350 MHz

100 MHz ~500 MHz

Sample rate

20 GSa/s

5 GSa/s(10 GSa/s ESR)

5 GSa/s(10 GSa/s ESR)

5 GSa/s

4 GSa/s

2 GSa/s

2 GSa/s

Analog channel

4

4

4/8

4

4

4

2/4

Memory depth

1 Gpts500 Mpts

500 Mpts

250 Mpts

400 Mpts

200 Mpts

200 Mpts

Waveform update
Rate

1,100,000 wfm/s

750,000 wfm/s

750,000 wfm/s

500,000 wfm/s

890,000 wfm/s

500,000 wfm/s

500,000 wfm/s

Protocol analysis

Standard: I2C, SPI, UART, CAN, LIN
Optional: CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT, Manchester (decode only), ARINC429(only SDS7000A, SDS6000A, SDS5000A,
SDS3000X HD)

Sequence

Yes

Yes

Yes

Yes

Yes

Yes

Yes

History

Yes

Yes

Yes

Yes

Yes

Yes

Yes

Math traces

4

4

4

2

4

2

2

FFT points

32 Mpts

8 Mpts8 Mpts2 Mpts4 Mpts2 Mpts2 MptsSearch andNavigate

Yes

Yes

Yes

Yes

Yes

Yes

Yes

DVM

Yes

Yes

Yes

Yes

Yes

Yes

Counter

Yes

Yes

Yes

Yes

Yes

Yes

Yes

Histogram

Yes

Yes

Yes

Yes

Yes

Yes

Yes

Bode plot

Yes

Yes

Yes

Yes

Yes

Yes

Yes

Power analysis

Yes

Yes

Yes

Yes

Yes

Yes

Yes

Eye/Jitter analysis

Yes

Yes

Yes

Compliance Test

Yes

Digital channels

Yes

Yes

Yes

Yes

Yes

Yes

Yes

AWG

50 MHz25 MHz25 MHz25 MHz50 MHz25 MHz50 MHzZone Trigger

Yes

Yes

Yes

Yes

Yes

Yes

Yes

Webserver

Yes

Yes

Yes

Yes

Yes

Yes

Yes

HDMI output

Yes

Yes

Yes

Porbe Adapters

Yes

Yes

Yes

Yes

Yes

Screen15.6’’ touch

12.1’’ touchNone

10.1’’ touch10.1” touch

10.1’’ touch

10.1’’ touch

www.siglent.com 55

SDS2000X-E

SDS1000X HD

SDS800X HD

SDS1000X-E

SDS1104X-U

SDS1000CML+

Bandwidth

200 MHz ~ 350 MHz

100 MHz ~200 MHz

70 MHz ~200 MHz

100 MHz ~ 200 MHz

100 MHz70 MHz ~ 150 MHz

Sample rate

2 GSa/s

2 GSa/s

2 GSa/s

1 GSa/s

1 GSa/s

1 GSa/s

Analog channel

2

2/4

2/4

2/4

4

2

Memory depth

28 Mpts

100 Mpts

100/50 Mpts

14 Mpts

14 Mpts

2 MptsWaveform update
Rate

400,000 wfm/s

500,000 wfm/s

500,000 wfm/s

400,000 wfm/s

400,000 wfm/s

Protocol analysis

I2C, SPI, UART, CAN,
LIN

I2C, SPI, UART,
CAN, LIN, CAN
I2C, SPI, UART, CAN, LIN
FD(Decode Only),
FlexRay(Decode Only)

Sequence

Yes

Yes

Yes

Yes

Yes

History

Yes

Yes

Yes

Yes

Yes

Math traces

1

4

4

1

1

FFT points

1 Mpts2 Mpts2 Mpts1 Mpts

128 kpts

Search andNavigate

Yes

Yes

Yes

Yes

Yes

Yes

Yes

Yes

Yes

Yes

Yes

1

DVM
Counter
Histogram
Bode plot

Yes

Power analysis

Yes

Eye/Jitter analysis
Digital channels

Yes

Yes

Yes

Yes

AWG

25 MHz25 MHz25 MHz25 MHz

Yes

Yes

Yes

Yes

7’’ LCD

10.1” touch

7’’ touch

7’’ LCD

Zone Trigger
Webserver
HDMI output
Porbe Adapters
Screenwww.siglent.com 56

7’’ LCD

7’’ LCD※

Arbitrary Waveform Generator ※SDG7000A

SDG6000X

SDG2000X

SDG1000X

SDG800

Bandwidth

350/500 MHz, 1 GHz

200/300/500 MHz

40/80/120 MHz

30/60 MHz

5/10/30 MHz

Number of channels

2 Differential/
Single-ended

2 Single-ended

2 Single-ended

2 Single-ended

1 Single-ended

Output range± 24 V (48 V)

±10V

±10V

±10V

±10V

Digital bus(Optional)

16-bit, LVTTL or LVDS
output
Bit rate1μbps ~ 1 Gbps

Sampling rate

5 GSa/s

2.4 GSa/s(2X Interpolation)

1.2 GSa/s(4X Interpolation)

150 MSa/s

125MSa/s

Vertical resolution

14-bit16-bit16-bit14-bit14-bitArbitrary waveform
length24 pts ~ 512 Mpts/ch

2 ~ 20 Mpts

8 ~ 8 Mpts

16 kpts

16 kpts

Modulation types

AM, FM, PM, PWM, FSK,
PSK, ASK, QAM

AM,FM,PM,ASK,FSK,PSK,
PWM, QAM

AM,FM,PM,ASK,FSK,PSK,
PWM

AM, DSB-AM, FM,PM,
FSK, ASK, PSK, PWM

AM, DSB-AM, FM, PM,
FSK, ASK, PWM

Harmonic output

16

10

10

16

Sweep & Burst

Yes

Yes

Yes

Yes

Yes

IQ Signal Generator

Yes

Yes

PRBS Generator

Yes

Yes

Display

5’’ touch screen ,800*480

4.3” touch screen,
480*272

4.3” touch screen,
480*272

4.3” LCD, 480*272

3.5’’ LCD, 320*240

www.siglent.com 57

※Power Supply ※SPS5000X

SPD3000X

SPD3303C

SPD1000X

Output Channel

1/2/3

3

3

1

Max. Voltage

40/50/80/160 V

32 V

32 V

16/30 V

Max. Current

7.5/15/22.5/30/45/60/90 A

3.2 A

3.2 A

5/ 8 A

Max. Power180/360/720/1080 W

220 W

220 W

128/150 W

Resolution

1 mV/1 mA

1 mV/1 mA

10 mV/10 mA

1 mV / 1 mA

Screen2.4’’ OLED

4.3’’ LCD

LED

2.8’’ LCD

※ DC Electronic Load ※SDL1020X

SDL1020X-E

SDL1030X

SDL1030X-E

Min. readback resolution

0.1 mV, 0.1 mA

1 mV, 1 mA

0.1 mV, 0.1 mA

1 mV, 1 mA

Input power

200 W

Input current

30 A

Input voltage

150 V

CC Dynamic mode frequency

25 kHzCurrent slew rate

0.001 A/us~2.5 A/us

Display

3.5 inch TFT-LCD display

300 W※ Digital Multimeter ※SDM3045X

SDM3055

SDM3065X

Reading resolution

4 1/2

5 1/2

6 1/2

DC voltage

600 mV ~ 1000 V

200 mV ~ 1000 V

200 mV ~ 1000 V

AC voltage

600 mV ~ 750 V

200 mV ~ 750 V

200 mV ~ 750 V

DC current

600 μA ~ 10 A

200 μA ~ 10 A

200 μA ~ 10 A

AC current

60 mA ~ 10 A

20 mA ~ 10 A

200 μA ~ 10 A

Scanner card

Not support

SupportSupport

Display

4.3” TFT-LCD, 480*272

www.siglent.com 58

Service
Service Promise:
Since the date of purchase, we offer three year’s warranty for the main unit:• During the warranty period, if the products cause any hardware or software failure because of the quality, Siglent's after-sales service centeror Siglent's designated maintenance points will offer the maintenance of the fault products for the user.
• Because of improper use or any other artificial reason, the damage won’t be included in the free maintenance.

1. Extension after-sales service
Extension service is based on the main unit (not including accessories) as an object. During the extension service, Siglent still offer freemaintenance after the standard warranty period.
1.1 Three advantages:• Guarantee investment. To extend the life cycle of the products.
• Save money. To prevent the high cost of maintenance after the warranty period.
• Avoid the repeated investment. To prevent buying new equipments because it can’t be repaired after the warranty period.
1.2 The content of the extension service
You can buy the following extension service according to your demand:
Solution

Viability

ES4

One year after the warranty period

Instruction
According to the service terms, Siglent will offer
another one year for the after-sales maintenanceservice
According to the service terms, Siglent will offer

ES5

Two years after the warranty period

another two years for the after-sales maintenanceservice

2. Calibration services
After long-term use, oscilloscope will cause the deviation of measured value and waveform display, because of its work temperature andhumidity. Siglent will restore the original performance and accuracy of factory setting to calibrate the deviation.
• Eliminate the error of measurement• Restore the original performance and accuracy of the factory setting to the “new” state• The upgrade of the firmware and the software• Make the instruments comply with the standard of the ISO9001 quality management process• Traceable calibration certificateswww.siglent.com 59

Follow us on Facebook:SiglentTech
modification date：2024-05

Headquarters:
SIGLENT Technologies Co., Ltd
Add: Bldg No.4 & No.5, Antongda Industrial
Zone, 3rd Liuxian Road, Bao'an District,
Shenzhen, 518101, China
Tel: + 86 755 3688 7876
Fax: + 86 755 3359 1582
Email: sales@siglent.com
Website: int.siglent.com

USA:
SIGLENT Technologies America, Inc
6557 Cochran Rd Solon, Ohio 44139
Tel: 440-398-5800
Toll Free: 877-515-5551
Fax: 440-399-1211
Email: info@siglent.com
Website: www.siglentna.com

Every Bench. Every Engineer. Every Day.

Europe:SIGLENT Technologies Germany GmbH
Add: Staetzlinger Str. 70
86165 Augsburg, Germany
Tel: +49(0)-821-666 0 111 0
Fax: +49(0)-821-666 0 111 22
Email: info-eu@siglent.com
Website: www.siglenteu.com