SDS7000A Series
Digital Storage
Oscilloscope
Data Sheet
2022.02

EN01B

Key Features
4 analog channels, up to 4 GHz bandwidth with up to
20GSa/s sample rate

SDS7404A H1212-bit ADC

SDS7304A H12

SPO technology

Low background noise: 220 μVrms @ 4 GHz bandwidth

Waveform capture rates up to 1,000,000 wfm/s

Supports 256-level intensity grading and color
temperature display modes

500 Mpts/ch standard, 1 Gpts/ch optional



Digital trigger system

Intelligent trigger: Edge, Slope, Pulse, Window, Runt,
Interval, Dropout, Pattern, Qualified, Nth edge,
Setup/hold, Delay and Video (HDTV supported). Zone
Trigger simplifies advanced triggering

Product Overview

Serial bus triggering and decoder, supports protocols I2C,
SPI, UART, CAN, LIN, CAN FD, FlexRay, I2S, MIL-STD1553B, SENT, Manchester, ARINC429 and USB 2.0

SIGLENT’s SDS7000A series Digital Storage Oscilloscopes
are available in bandwidths of 4 GHz and 3 GHz, have 12-bit
ADCs with sample rate up to 20 GSa/s, maximum record lengthof 1Gpts/ch, and display up to 4 analog channels + 16 digitalchannels for high performance mixed signal analysis.

Segmented acquisition (Sequence) mode, dividing the
maximum record length into multiple segments (up to124,000), according to trigger conditions set by the user,
with a very small dead time between segments to capture
the qualifying event

The SDS7000A series employs Siglent’s SPO technology
with a maximum waveform capture rate of up to 1,000,000
wfm/ s, 256-level intensity grading display function plus a color
temperature display mode. It also employs an innovative digitaltrigger system with high sensitivity and low jitter. The trigger
system supports multiple powerful triggering modes including
serial bus triggering. Tools such as History waveformrecording, Search and Navigate functions, Signal Scan, Mask
Test, Bode Plot, Power Analysis, Eye/Jitter Analysis andCompliance Test allow for extended waveform records to be
captured, stored, and analyzed. An impressive array ofmeasurement and math capabilities, options for a 50 MHz
arbitrary waveform generator, as well as serial decoding arealso features of the SDS7000A .
The large 15.6’’ HD display capacitive touch screen supports
multi-touch gestures, with the addition of user-friendly UIdesign, can greatly improve the operational efficiency. It also
supports mouse control, and remote web control over LAN.

History waveform record (History) function, the maximum
recorded waveform length is 124,000 frames
Automatic measurements on 50+ parameters, supports
statistics
withhistogram, track, trend, Gating
measurement, and measurements on Math, History andMemory traces
4 Math traces (32 Mpts FFT, Filter, addition, subtraction,
multiplication, division, integration, differential, squareroot, etc.), supports formula editor
Abundant data analysis functions such as Search,
Navigate, SignalScan, Digital Voltmeter, Counter,
Waveform Histogram, Bode plot, Power Analysis,
Eye/Jitter Analysis and Compliance Test
High Speed hardware-based Average, Hi-Res; High
Speed hardware-based Mask Test function, with Mask
Editor tool for creating user-defined masks
16 digital channels
Built-in 50 MHz waveform generator
Large 15.6” HD TFT-LCD display with 1920 * 1080
resolution; Capacitive touch screen supports multi-touchgesturesInterfaces include: 4x USB Host 3.1 Gen 1，2x USB 3.0
Host，USB 2.0 Device，2x 1000M LAN，DVI-D，DP 1.2，
HDMI 1.4，Audio, External Triger In, Aux Out (Pass/Fail,
Trigger Out), 10 MHz In，10 MHz Out
Built-in web server supports remote control over the LAN
port using a web browser. Supports SCPI remote control
commands. Supports external mouse and keyboard

WWW.SIGLENT.COM

1

SDS7000A Series Digital Storage Oscilloscope

Models and Key Specifications
Model

SDS7404A H12

Analog channels
Bandwidth
Sample rate
(Max.)
VerticalResolution
Memory depth(Max.)
Waveform capture
rate (Max.)

4 + EXT
4 GHz
20 GSa/s (dual-channel)
10 GSa/s (3 or 4 channels)
12-bitUp to 16-bit in ERES modeStandard: 500 Mpts/ch
Optional: 1 Gpts/ch in dual-channel mode

SDS7304A H12

Trigger type

Edge, Slope, Pulse width, Window, Runt, Interval, Dropout, Pattern, Video, Qualified, Nth edge,
Setup/hold, Delay, Serial

Serial trigger anddecodeStandard: I2C, SPI, UART, CAN, LIN
Optional: CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT, Manchester (decode only), ARINC429, USB 2.0(decode only)

Measurement

50+ parameters，statistics, histogram, trend, and track supported

3 GHz

1,000,000 wfm/s

4 traces
MathData analysis
Digital channel
Waveform
generator(optional)
Processor System

32 Mpts FFT, +, -, x, ÷, ∫dt, d/dt, √, Identity, Negation, Absolute, Sign, ex, 10x, ln, lg, Interpolation, MaxHold,
MinHold, ERES, Average, Filter. Supports formula editor
Search, Navigate, History, Mask Test, Digital Voltmeter, Counter, Waveform Histogram, Bode plot andPower Analysis, Eye/Jitter Analysis, SignalScan, Compliance Test (USB 2.0, 100Base-TX, 1000Base-T,
100Base-T1, 1000Base-T1)
16-channel; maximum sample rate up to 1 GSa/s; record length up to 50 Mpts
Builit-in, frequency up to 50 MHz, 125 MSa/s sample rate, 16 kpts waveform memory
Intel Core i3-8100 or better，32GB memory，250GB storage，Linux operating system
I/O: 4x USB Host 3.1 Gen 1，2x USB 3.0 Host，USB 2.0 Device (USBTMC)，2x 1000M LAN (VXI11+SCPI， Telnet (5024)+SCPI，Socket (5025)+SCPI，LXI，WebServer)

I/O

Display: 1x DVI-D: up to 1920x1200 @ 60Hz; 1x DP 1.2: up to 4096x2304 @ 60Hz; 1x HDMI 1.4: up to4096x2160 @ 60Hz
Audio: Mic input, Audio Output
Others: External Trigger In, Aux Out (TRIG OUT，PASS/FAIL)，10 MHz In，10 MHz Out

Probe (Standard)
Display2

500 MHz, 1 probe supplied for each channel
15.6” HD TFT-LCD with capacitive touch screen (1920*1080)

WWW.SIGLENT.COM

SDS7000A Series Digital Storage Oscilloscope

Functions & Characteristics
12-bit High Resolution

8-bit

12-bit12-bit resolution shows you more details and less noise on the waveform.

Upgraded processor system

Processor fully upgraded from the embedded ARM processor to
the X86 processor, has greatly improved the system response
speed and the speed of measurement, calculation, andanalysis, presenting more possibilities for the expansion ofsoftware analysis functions in the future.

Excellent User Interface and User Experience

WWW.SIGLENT.COM



15.6” HD display with 1920*1080 resolution

Capacitive touch screen, supporting multi-touchgestures, can move or scale the waveform traces
quickly by finger-touch movements, which greatly
improves the operational efficiency



Built-in WebServer supports remote control on a
web page over LAN

Supports external mouse and keyboard

3

SDS7000A Series Digital Storage OscilloscopeHigh Waveform Update Rate

Deep Record LengthWith a waveform update rate of up to 1,000,000 wfm/s, theoscilloscope can easily capture unusual or low-probability
events. In Sequence mode, the waveform capture rate canreach 1,100,000 wfm/s

Using hardware-based Zoom technique and record length of up
to 1 Gpts, users can select a slower timebase withoutcompromising the sample rate, and then quickly zoom in tofocus on the area of interest

Multiple Trigger Functions

Zone Trigger

Edge, Slope, Pulse, Video, Windows, Runt, Interval, Dropout,
Pattern, Qualified, Nth edge, Setup/hold, Delay and serial
trigger

Zone Trigger is available for advanced triggering. Combine
spatial triggering with common trigger modes to isolate signalsof interest

4

WWW.SIGLENT.COM

SDS7000A Series Digital Storage Oscilloscope

Advanced Math Function

Deep Memory FFT

In addition to the traditional (+, -, X, /) operations, FFT, Filter,
integration, differential, square root, and more are supported.
Formula Editor is available for more complex operations. 4 mathtraces are available.

FFT supports up to 32 Mpts operation. This provides highfrequency resolution with a fast refresh rate. The FFT function
also supports a variety of window functions so that it can adapt
to different spectrum measurement needs. Three modes(Normal, Average, and Max hold) can satisfy different
requirements for observing the power spectrum. Auto peakdetection and markers are supported.

Measurements of a Variety of Parameters

Parameter Statistics Function

Parameter measurements include 4 categories: horizontal,
vertical, miscellaneous, and CH delay providing a total of 50+
different types of measurements. Measurements can be
performed within a specified gate period. Measurements on
Math, Reference, and History frames are supported

WWW.SIGLENT.COM

Statistics show the current value, maximum value, minimum
value, standard deviation, and mean value of up to 12
parameters simultaneously. A histogram is available to show the
probability distribution of a parameter. Trend and Track areavailable to show the parameter value vs. time.
For horizontal parameters such as period, all results are
extracted from a frame, instead of just calculating the first one.
This accelerates statistics on horizontal measurements andenables distribution observation in a frame using Histogram and
Track

5

SDS7000A Series Digital Storage Oscilloscope

History Mode

Sequence Mode

History function can record up to 124,000 frames of waveforms.
The recording is executed automatically so that the customer
can playback the history waveforms at any time to observe
unusual events and quickly locate the area of interest using the
cursors or measurements. The failed frames of the Mask Test
can be stored as history

Segmented memory collection will store the waveform into
multiple memory segments (up to 124,000) and each segment
will store a triggered waveform as well the dead time
information. The interval between segments can be as small as
0.9 μs. All of the segments can be played back using the History
function

Search and Navigate

Serial Bus Decode

The oscilloscope can search events specified by the user in aframe. Events flagged by the Search can be recalled
automatically using Navigate. It can also navigate by time (delay
position) and history frames

Display the decoded characters through the events list. Bus
protocol information can be quickly and intuitively displayed intabular form. I2C, SPI, UART, CAN, LIN, CAN FD, FlexRay, I2S,
MIL-STD-1553B, SENT, Manchester , ARINC429 and USB 2.0
are supported

6

WWW.SIGLENT.COM

SDS7000A Series Digital Storage Oscilloscope

Hardware-based High Speed Mask Test Function

The oscilloscope utilizes a hardware-based Mask Test function,
performing up to 80,000 Pass / Fail decisions each second. It iseasy to generate user-defined test templates to provide trace
mask comparisons, making it suitable for long-term signal
monitoring or automated production line testing

Built-in Mask Editor application helps to create custom masks

Eye/Jitter Analysis

Supports eye diagram and jitter analysis/measurement. It can automatically extract the embedded reference clock from serial dataand create the eye diagram. Measurement on multiple eye/jitter parameters is provided. Mask test on eye diagrams is supported

WWW.SIGLENT.COM

7

SDS7000A Series Digital Storage Oscilloscope

Compliance Test（Optional）
USB 2.0, 100Base-TX, 1000Base-T, 100Base-T1, 1000BaseT1 protocol conformance testing are available. When the user
sets up the environment according to the prompts, by using therelated test fixture, the oscilloscope and related instruments canbe automatically set up and related measurement, calculation,
decoding and other functions will be used for testing, helping theuser to complete each test project quickly and efficiently, andreports are generated automatically.

Bode PlotPower Analysis (Optional)

The oscilloscope can control the Built-in waveform generator,
SIGLENT isolated USB AWG module or a stand-alone
SIGLENT SDG generator, to scan the amplitude and phasefrequency response of the DUT, and display the data as a Bode
Plot. This makes it possible to replace expensive network
analyzers in some applications

The Power Analysis option provides a full suite of power
measurements and analysis, which greatly improve themeasurement efficiency in switching power supplies and powerdevices design

Digital Channels / MSO

Builit-in 50 MHz Function/Arbitrary Waveform
Generator (Optional)

Four analog channels plus 16 digital channels enable users toacquire and trigger the waveforms then analyze the pattern,simultaneously with one instrument

The oscilloscope can control the built-in waveform generator to
output waveform with up to 50 MHz frequency and ±3 V
amplitude. Six basic waveforms plus multiple types of arbitrary
waveforms are built-in

8

WWW.SIGLENT.COM

SDS7000A Series Digital Storage Oscilloscope

5 GHz Active Differential Probe

The SAP5000D differential probe is provided with 5GHz
bandwidth, 80 ps rise time, 400 fF differential input capacitance,and 10:1 attenuation ratio

Complete Connectivity


2x USB 3.0 Host， 4x USB Host 3.1 Gen 1，USB 2.0 Device（USBTMC）, 2x 1000M
LAN（VXI-11+SCPI， Telnet（5024）+SCPI，Socket（5025）+SCPI，LXI，
WebServer），



1x DVI-D: up to 1920x1200 @ 60Hz，1x DP 1.2: up to 4096x2304 @ 60Hz，1x HDMI 1.4:
up to 4096x2160 @ 60Hz

Mic input, Audio Output

External Trigger In，Aux Out (TRIG OUT，PASS/FAIL), 10 MHz In，10 MHz Out

WWW.SIGLENT.COM

9

SDS7000A Series Digital Storage Oscilloscope

Specifications
All specifications are not guaranteed unless the following conditions are met:
• The oscilloscope calibration period is current• The oscilloscope has been working continuously for at least 30 minutes at the specified temperature (18℃ ~ 28℃ )Acquire (analog)
Sample rate
Memory depth *2*3
Real time signal
processing depth
Waveform update
rate
Intensity grading
Peak detect
Average
ERES
Sequence
History
Interpolation

20 GSa/s (dual-channel) *1
10 GSa/s (3 or 4 channels) *1
Standard: 500 Mpts/ch
Optional: 1 Gpts/ch in dual-channel mode
Measure, math, decode, analysis: 100 Mpts/ch max.
1,000,000 wfm/s, 1,100,000 wfm/s in sequence mode
256-level
100 ps4, 16, 32, 64, 128, 256, 512, 1024, 2048, 4096, 8192
Enhanced bit: 0.5, 1, 1.5, 2, 2.5, 3, 3.5, 4 bit
Up to 124,000 segments, interval between triggers = 0.9 μs min.
Up to 124,000 frames
sinx/x, x* 1：dual-channel: C1/C2 are not both active, and C3/C4 are not both active* 2：In Average and Hi-Res modes，the memory depth is 25 Mpts/ch* 3：When digital channels are active，the memory depth is 50 Mpts/ch

Vertical(analog)

SDS7404A H12

SDS7304A H12ChannelBandwidth(-3dB) @ 50Ω

4 + EXT
4 GHz

3 GHz

Rise time@50Ω

116 ps typical< 125 ps

130 ps typical< 150 ps

500 MHz

500 MHz

300 MHz

300 MHz

Bandwidth(-3dB) @ 1 MΩ,with probe
Bandwidth(-3dB) @ 1 MΩ,with external 50Ω
Resolution

Bandwidth in
ERES mode(typical)

12-bitEnhanced bits:
0.5: 0.25*Sample rate, up to the analog bandwidth
1: 0.115*Sample rate, up to 2.3 GHz, limited by the analog bandwidth
1.5: 0.055*Sample rate, up to 1.1 GHz, limited by the analog bandwidth
2: 0.028*Sample rate, up to 560 MHz
2.5: 0.014*Sample rate, up to 280 MHz
3: 0.007*Sample rate, up to 140 MHz
3.5: 0.0035*Sample rate, up to 70 MHz
4: 0.0017*Sample rate, up to 34 MHz

Noise floor (rms,50Ω,typical)≤ 5 mV/div

220 μV

200 μV

10 mV/div

237 μV

214 μV

20 mV/div

280 μV

255 μV

50 mV/div

635 μV564 μV

100 mV/div

1.01 mV

935 μV

200 mV/div
500 mV/div
1 V/div
ENOB*1 (typical)
Range

3.06 mV
6.84 mV
10.65 mV
7.3-bit
8 divisions

2.68 mV
5.89 mV
9.58 mV
7.5-bit10

WWW.SIGLENT.COM

SDS7000A Series Digital Storage Oscilloscope
Vertical scale(probe 1X)
DC gain accuracy
Offset accuracy
Offset range(probe 1X)
Bandwidth limit
Low frequency
response (ACcoupling -3 dB)
Coupling
Impedance
Max. Input
voltage
SFDR
CH to CH
Isolation (@50Ω)Probe Attenuation

1 MΩ: 1 mV/div – 10 V/div
50 Ω: 1 mV/div – 1 V/div
1 mV/div ~ 4.95 mV/div: ±1.5%5 mV/div ~ 10 V/div: ±1% max. ±0.5% typical± (1% of the offset setting + 0.5% of full scale + 0.02% of max offset + 1mV)
1MΩ: 1 mV/div ~ 5 mV/div: ±1.6 V; 5.1 mV/div ~ 10 mV/div: ±4 V; 10.2 mV/div ~ 20 mV/div:±8 V; 20.5
mV/div ~ 100 mV/div: ±16 V；102 mV/div ~ 200 mV/div: ±80 V; 205 mV/div ~ 1 V/div: ±160 V; 1.02 V/div ~10 V/div: ±400 V
50Ω: 1 mV/div ~ 5 mV/div: ±1.6 V; 5.1 mV/div ~ 10 mV/div: ±4 V; 10.2 mV/div ~ 20 mV/div:±8 V; 20.5
mV/div ~ 1 V/div: ±10 V
25 MHz, 200 MHz, Custom
6 Hz (typical)
DC, AC, GND(1 MΩ ± 2%) || (15 pF ± 3 pF)
50 Ω: 50 Ω ± 2%
1 MΩ ≤ 400 Vpk(DC + AC), DC ~ 10 kHz
50 Ω ≤ 5 Vrms, ± 10V Peak
≥ 45dBc
60 dB up to 500 MHz
30 dB up to 4 GHz
1X, 10X, 100X, custom*1: 50 Ω, 50 mV/div, 20 GSa/s, -1dBFS/47.999 MHz input

Horizontal
Time scale
Range
Display mode
Roll mode
Skew (C1~C4)
Time base
Accuracy

0.05 ns/div – 1000 s/div
10 divisions
Y-T, X-Y, Roll
≥ 50 ms/div< 100 psStandard (TCXO): ±2 ppm initial (0~50℃); ±0.5 ppm 1st year aging; ±3 ppm 20-year aging
Optional (OCXO)：±100 ppb initial (25℃)；±1 ppb over temperature (0~50℃)；±50 ppb 1st year aging

TriggerMode

Auto, Normal, Single
Internal: ±4.5 div from the center of the screen

Level

Ext Trigger Channel
input voltage
Hold off range

Coupling

EXT: ± 0.61 V
EXT/5: ± 3.05 V
1 MΩ ≤ 42 Vpk
50 Ω ≤ 5 Vrms
By time: 4 ns ~ 30 s (4 ns step)
By event: 1 ~ 108
C1~C4
DC: Passes all components of the signal
AC: Blocks DC components and attenuates signals below 15 Hz
LFRJ: Attenuates the frequency components below 2.4 MHz
HFRJ: Attenuates the frequency components above 1.3 MHz
Noise RJ: Increases the trigger hysteresis
EXT
DC: Passes all components of the signal
AC: Blocks DC components and attenuates signals below 10 Hz
LFRJ: Attenuates the frequency components below 500 kHz
HFRJ: Attenuates the frequency components above 1.8 MHz

Accuracy (typical)

C1～C4: ±0.2 div
EXT: ±0.3 div
C1 ～ C4 (50Ω):Sensitivity

C1 ～ C4 (1MΩ):EXT:

WWW.SIGLENT.COM> 10 mV/div≤ 10 mV/div> 5 mV/div≤ 5 mV/div

Noise RJ = OFF
0.8 div
2.4 div
0.5 div
1.5 div

Noise RJ = ON
1.0 div
2.6 div
0.7 div
1.7 div

200 mVpp, DC ～ 10 MHz
11

SDS7000A Series Digital Storage Oscilloscope
300 mVpp, 10 MHz ～ bandwidth（300 MHz）
EXT/5:

1 Vpp, DC ～ 10 MHz
1.5 Vpp, 10 MHz ～ bandwidth（300 MHz）

Jitter

C1～C4:< 9 ps RMS (typical) for ≥ 300 MHz sine and ≥ 6 divisions peak to peak amplitude for vertical gain settingsfrom 2.5 mV/div to 10 V/div< 5 ps RMS (typical) for ≥ 500 MHz sine and ≥ 6 divisions peak to peak amplitude for vertical gain settingsfrom 2.5 mV/div to 10 V/div
EXT: < 50 ps rms

Displacement

Pre-Trigger: 0 ~ 100% memory
Delay-Trigger: 0 ~ 10,000 div
Up to 2 zones

Zone
Edge Trigger
SourceSlope
Slope Trigger
SourceSlope
Limit range

Source: C1~C4
Property: Intersect, Not Intersect
C1~C4/EXT/(EXT/5)/AC Line/D0~D15
Rising, Falling, Rising & Falling
C1~C4
Rising, Falling<, >, in range, out of range

Time range

2 ns ～ 20 s, Resolution = 0.2 ns

Pulse Width Trigger
SourcePolarity
Limit range

C1~C4/D0~D15+wid, -wid<, >, in range, out of range

Time range

2 ns ～ 20 s, Resolution = 0.2 ns

Video Trigger
SourceStandard
Synchronization
Trigger Condition
Window Trigger
SourceWindow typeInterval Trigger
SourceSlope
Limit range

C1~C4
NTSC, PAL, 720p/50, 720p/60, 1080p/50, 1080p/60, 1080i/50, 1080i/60, Custom
Any, SelectLine, Field
C1~C4
Absolute, Relative
C1~C4/D0~D15
Rising, Falling<, >, in range, out of range

Time range

2 ns ～ 20 s, Resolution = 0.2 ns

Dropout Trigger
SourceTimeout type
Slope

C1~C4/D0~D15
Edge, State
Rising, Falling

Time range

2 ns ～ 20 s, Resolution = 0.2 ns

Runt Trigger
SourcePolarity
Limit range

C1~C4
Positive, Negative<, >, in range, out of range

Time range

2 ns ～ 20 s, Resolution = 0.2 ns

Pattern Trigger
SourcePattern Setting
Logic
Limit range

C1~C4/D0~D15
Don’t Care, Low, HighAND, OR, NAND, NOR<, >, in range, out of range

Time range

2 ns ～ 20 s, Resolution = 0.2 ns

Qualified Trigger
Type
Qualified Source

State, State with Delay, Edge, Edge with Delay
C1~C4/D0~D15

12WWW.SIGLENT.COM

SDS7000A Series Digital Storage Oscilloscope
Edge Trigger Source
Nth Edge Trigger
SourceSlope
Idle time

C1~C4/D0~D15

Edge Number

1 ～ 65535

Delay Trigger
Source A
Source B
Slope
Limit range

C1~C4/D0~D15
C1~C4/D0~D15
Rising, Falling<, >, in range, out of range

Time range

2 ns ～ 20 s, Resolution = 0.2 ns

Serial Trigger
SourceC1~C4/D0~D15
Rising, Falling
8 ns ～ 20 s, Resolution = 0.2 ns

I2 C
SPI
UART
CAN
LIN

C1~C4/D0~D15
Standard: I2C, SPI, UART, CAN, LIN
Optional: CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT, ARINC429
Type: Start, Stop, Restart, No Ack, EEPROM, Address & Data, Data Length
Type: Data
Type: Start, Stop, Data, Parity Error
Type: All, Remote, ID, ID+Data, Error
Type: Break, Frame ID, ID+Data, Error

CAN FD（Optional）

Type: Start, Remote, ID, ID+Data, Error

FlexRay（Optional）

Type: TSS, Frame, Symbol, Errors

I2S（Optional）

Type: Data, Mute, Clip, Glitch, Rising Edge, Falling Edge

MIL-STD-1553B（Optional）

Type：Transfer, Word, Error, Timing

SENT（Optional）

Type: Start, Slow channel, Fast channel, Error

ARINC429 (Optional)

Type: Word Start, Word End, Label, Label+Data, Error, Any Bit, Any Bit of 0, Any Bit of 1

ProtocolSerial Decoder
Decoders

2

Threshold-4.1 ～ 4.1 div

List

1 ～ 7 lines

Decoder type
I2 C
SourceSignalAddress
SPI
SourceSignalEdge SelectChip selectBit Order
UART
SourceSignalData Width
Parity Check
Stop Bit
Idle Level
Bit Order
CAN
SourceLIN
LIN Version
SourceBaud Rate

Full duplex
C1~C4/D0~D15
SCL, SDA
7-bit, 10-bit
C1~C4/D0~D15
CLK, MISO, MOSI, CS
Rising, Falling
Active high, Active low, Clock timeout
LSB, MSB
C1~C4/D0~D15
RX, TX
5-bit, 6-bit, 7-bit, 8-bit
None, Odd, Even, Mark, Space
1-bit, 1.5-bit, 2-bit
Low, High
LSB, MSB
C1~C4/D0~D15
Ver 1.3, Ver 2.0
C1~C4/D0~D15
600 bps, 1200 bps, 2400 bps, 4800 bps, 9600 bps, 19200 bps, Custom

CAN FD（Optional）
WWW.SIGLENT.COM

13

SDS7000A Series Digital Storage Oscilloscope
SourceNominal Baud Rate
Data Baud Rate

C1~C4/D0~D15
10 kbps, 25 kbps, 50 kbps, 100 kbps, 250 kbps, 1 Mbps, Custom
500 kbps, 1 Mbps, 2 Mbps, 5 Mbps, 8 Mbps, 10 Mbps, Custom

FlexRay（Optional）
SourceBaud Rate

C1~C4/D0~D15
2.5 Mbps, 5 Mbps, 10 Mbps, Custom

I2S（Optional）
SourceSignalAudio Variant
Start Bits
Data Bits

C1~C4/D0~D15
BCLK, WS, DATA
Audio-I2S, Audio-LJ, Audio-RJ
0~31
1~32

MIL-STD-1553B（Optional）
SourceC1~C4

SENT（Optional）
SourceC1~C4/D0~D15

Manchester（Optional）
SourceBaud Rate

C1~C4
500 bps~5 Mbps

ARINC429（Optional）
SourceBaud Rate

C1~C4
12.5 kbps~100 kbps, tolerance 1%~20%

Word format

L/SDI/D/SSM，L/D/SSM，L/D

USB 2.0（Optional）
SourceData type

Full speed/Low speed: C1~C4/D0~D15
High speed: C1~C4
Low speed (1.5 Mbps), Full speed (12 Mbps), High speed (480Mbps)

Measurement
Automatic Measurement
SourceC1~C4, D0~D15, Z1~Z4, F1~F4, M1~M4, History
Mode
Simple, Advanced
ScreenRange
Gated: inside screen, definable with separate Gate cursors
Custom Threshold
Upper, Middle, Lower
No. of Measurements
Display 12 measurements at the same time (Display mode = M2)
Max, Min, Pk-Pk, Top, Base, Amplitude, Mean, Cycle Mean, Stdev, Cycle Stdev, RMS, Cycle RMS,
Vertical Parameters
Median, Cycle Median, FOV, FPRE, ROV, RPRE, Level@Trigger
Period, Frequency, Time@max, Time@min, +Width, -Width, 10-90%Rise time, 90-10%Fall time, Rise
Horizontal Parameters
time, Fall time, +Burst Width, -Burst Width, +Duty Cycle, -Duty Cycle, Delay, Time@Middle, CycleCycle jitter+Area@DC, -Area@DC, Area@DC, Absolute Area@DC, +Area@AC, -Area@AC, Area@AC,
Miscellaneous
Absolute Area@AC, Cycles, Rising Edges, Falling Edges, Edges, Positive pulses, Negative pulses,
Parameters
Positive Slope, Negative Slope
Delay Parameters
Phase, FRFR, FRFF, FFFR, FFFF, FRLR, FRLF, FFLR, FFLF, Skew
Current, Mean, Min, Max, Sdev, Count;
Statistics
Histogram, Trend, Track
Statistics Count
Unlimited, 1~1024
Statistics Count in one
Up to 100,000
frame
Cursors
SourceC1~C4, Z1~Z4, D0~D15, F1~F4, M1~M4, Histogram
Manual : Time X1, X2, (X1-X2), (1/ΔT); Vertical Y1, Y2, (Y1-Y2)
Type
Track: Time X1, X2, (X1-X2)
Measure: indicates the measurement on specific parameter

Math
Trace
SourceOperation
14

F1, F2, F3, F4
C1~C4, F1~F4, M1~M4
FFT, +, -, x, ÷, ∫dt, d/dt, √, Identity, Negation, |x|, Sign, ex, 10x, ln, lg, Interpolation, Max hold, Min
hold, ERES, Average, Filter, Formula Editor
WWW.SIGLENT.COM

SDS7000A Series Digital Storage Oscilloscope

FFT

Length: 32 Mpts, 16 Mpts, 8 Mpts, 4 Mpts, 2 Mpts, 1 Mpts, 512 kpts, 256 kpts, 128 kpts, 64 kpts, 32
kpts, 16 kpts, 8 kpts, 4 kpts, 2 kpts
Window: Rectangular, Blackman, Hanning, Hamming, Flattop
Display: Full Screen, Split, Exclusive
Mode: Normal, Max hold, Average
Tools: Peaks, Markers

Analysis
SearchSourceMode
Copy setting
SignalScan
SourceMode
Copy setting
Navigate
Type
Mask Test

C1~C4, History
Edge, Slope, Pulse, Interval, Runt
Copy from trigger, Copy to trigger
C1~C4, F1~F4, M1~M4, D0~D15
Edge, Non-monotonic, Runt, Measure, Serial pattern, Bus pattern

Search event, Time, History frame

SourceC1~C4，Z1~Z4

Mask creating
Mask test speed
DVM
Source
Mode
Plot
Gate
Bode Plot
SourceAuto (Create mask), Customized (Mask Editor)
Up to 80,000 frames/s
C1~C4
DC mean, DC RMS, AC RMS, Peak-peak, Amplitude
Bar, Histogram, Trend
20 ms

C1~C4
Built-in waveform generator,
Supported signal
SAG1021I (Connection: USB),
sources
SDG series waveform generators (Connection: USB, LAN)
Sweep type
Simple, Vari-level
Mode: Linear, Logarithmic
Frequency
Range: 10 Hz ~ 120 MHz
MeasureUpper cutoff frequency, Lower cutoff frequency, Bandwidth, Gain margin, Phase margin
Power Analysis (optional)
Power quality, Current Harmonics, Inrush current, Switching loss, Slew rate, Modulation, Output
Measureripple, Turn on/turn off, Transient response, PSRR, Efficiency, SOA
Histogram
SourceC1~C4
Type
Horizontal, Vertical, BothCounter
SourceC1~C4Frequency resolution
7 digits
Totalizer
Counter on edges, supports Gate and Trigger
Eye Diagram (optional)
SourceC1~C4
Clock recovery
Constant frequency, PLL
MeasureEye height, “1”level, “0”level, Eye amplitude, Eye width, Eye crossing, Average power, Q factor, TIE
Mask Test
Supported
Jitter Analysis (optional)
SourceC1~C4
Clock recovery
Constant frequency, PLL
Period, Frequency, +Width, -Width, +Duty cycle, -Duty cycle, Cycle-cycle jitter, Cycle-cycle +width,
MeasureCycle-cycle -Width, Cycle-cycle +Duty cycle, Cycle-cycle -Duty cycle, Bit Rate, Unit interval
TIE, RJ, DJ, DCD, DDJ, PJ, TJ@BER
Jitter decomposition
Statistics: Histogram, Track, Spectrum

Compliance Test（Optional）
USB2.0Specification

WWW.SIGLENT.COM

USB 2.0 Electrical Compliance Test Specification, Version 1.07

15

SDS7000A Series Digital Storage Oscilloscope
Items

Ethernet
Specification
Items

Specification
Items

Specification
Items

Specification
Items

EL_1, EL_2, EL_3, EL_4, EL_5, EL_6, EL_7, EL_9, EL_21, EL_22, EL_23, EL_25, EL_27, EL_28,
EL_29, EL_31, EL_33, EL_34, EL_35, EL_38, EL_39, EL_40, EL_41, EL_42, EL_43, EL_44, EL_45,
EL_46, EL_47, EL_48, EL_55
100Base-TX
AOI Template, Peak Voltage (POS, NEG, Signal Amplitude Symmetry), Overshoot (POS, NEG),
Rise/Fall Times (POS Rise Time, Pos Fall Time, POS Rise/Fall Symmetry, NEG Rise Time, NEG Fall
Time, NEG Rise/Fall Symmetry, Overall Rise/Fall Symmetry), Duty Cycle Distortion, Peak to PeakTransmit Jitter, Return Loss (Transmitter Return Loss, Receiver Return Loss)
1000Base-T
No Disturber Peak Output Voltage (Point A, Point B, Difference A and B, Point C, Point D), No
Disturber Droop(Point G, Point J), No Disturber Templates(Point A, Point B, Point C, Point D, Point F,
Point H), No Disturber Transmitter Distortion(no TX_TCLK, with TX_TCLK), With Disturber Peak
Output Voltage (Point A, Point B, Difference A and B, Point C, Point D), With Disturber Droop (Point
G, Point J), With Disturber Templates (Point A, Point B, Point C, Point D, Point F, Point H), With
Disturber Transmitter Distortion (no TX_TCLK, with TX_TCLK), No TX_TCLK Master Jitter (Filtered,Unfiltered), No TX_TCLK Slave Jitter (Filtered, Unfiltered), Master JTXOUT, With TX_TCLK Master
Jitter (Filtered, Unfiltered), Slave JTXOUT, With TX_TCLK Slave Jitter (Filtered, Unfiltered), Return
Loss, Common-mode Output Voltage
100Base-T1
Transmitter Output Droop (Transmitter Output Droop(POS)/Transmitter Output Droop(NEG)), MasterTransmitter Clock Frequency And Timing Jitter (Master Transmitter Clock Frequency/MasterTransmitter Timing Jitter), TX_TCLK Frequency And Timing Jitter (TX_TCLK Frequency/TX_TCLKTiming Jitter), Transmitter Distortion, MDI Return Loss, MDI Mode Conversion Loss, Transmitter
Power Spectral Density And Peak Differential Output (Transmitter Power Spectral Density/Transmitter
Peak Differential Output), MDI Common Mode Emission
1000Base-T1
TX_TCLK125 Tests (TX_TCLK125 Frequency/Master TX_TCLK125 RMS Jitter/ Master TX_TCLK125
Peak-to-Peak Jitter/Slave TX_TCLK125 RMS Jitter/ Slave TX_TCLK125 Peak-to-PeakJitter),Transmit Clock Frequency And MDI Jitter (Transmit Clock Frequency(Master)/MDI Output RMS
Jitter(Master)/MDI Output Peak-to-Peak Jitter(Master)),Transmitter Distortion, MDI Return Loss, MDI
Mode Conversion Loss, Transmitter Power Spectral Density And Peak Differential Output(Transmitter
Power Spectral Density/Transmitter Peak Differential Output), Transmitter Output Droop(Transmitter
Output Droop(POS)/Transmitter Output Droop(NEG))

Digital Channels
Max. Sampling Rate
Memory Depth
Min. Detectable Pulse
Width
Level Group
Level Range
Logic TypeSkew

1 GSa/s
50 Mpts/ch
3.3 ns
D0~D7, D8~D15-10 V~10 V
TTL, CMOS, LVCMOS3.3, LVCMOS2.5, Custom
D0~D15: ±1 sampling interval
Digital to Analog: ± (1 sampling interval +1 ns)

Waveform Generator (optional)
ChannelsMax. Output Frequency
Sampling Rate
Frequency Resolution
Frequency Accuracy
Vertical Resolution
Amplitude Range
Waveforms
Output ImpedanceProtection
Sine
Frequency
Offset accuracy (10kHz)
Amplitude flatness
SFDR
Harmonic distortion
16

1
50 MHz
125 MSa/s
1 μHz±50 ppm
14 bit-1.5 V ~ +1.5 V (into 50 Ω)-3 V ~ +3 V (into High-Z)
Sine, Square, Ramp, Pulse, DC, Noise, 45 Arbitrary
50 Ω ± 2%
Over voltage protection, Current limit
1 μHz ～ 25 MHz±(1%*offset setting value +3 mVpp)±0.3 dB, compare to 10 kHz, 2.5 Vpp into 50 Ω
DC ~ 1 MHz-60 dBc
1 MHz ~ 5 MHz -55 dBc
5 MHz ~ 25 MHz -50 dBc
DC ~ 5 MHz-50 dBc
WWW.SIGLENT.COM

SDS7000A Series Digital Storage Oscilloscope
5 MHz ~ 25 MHz-45 dBc

Square/Pulse
Frequency

1 μHz ～ 10 MHz

Duty cycle

1% ～ 99%

Edge< 24 ns (10% ～ 90%)

Overshoot< 3% (typical，1 kHz, 1 Vpp)

Pulse width
Jitter (cycle-cycle)
Ramp
Frequency> 50 ns< 500 ps + 10 ppm

Linearity
Channels< 0.1% of Pk-Pk (typical, 1 kHz, 1 Vpp, 50% symmetry)

1 μHz ～ 300 kHz
0% ～ 100%

DC
Offset range
Accuracy
Noise
Bandwidth (-3 dB)
Arb
Frequency
Waveform memory
Sample rate
Wave import±1.5 V (into 50 Ω)±3 V (into Hi-Z)±(|setting value|*1% + 3 mV)>25 MHz
1 μHz ～ 5 MHz
16 kpts
125 MSa/s
From EasyWaveX, from U-disk, directly from waveform data of analog channels

Processor system
CPU
MemoryStorage
Operating system

Intel Core i3-8100 or better
32 GB DDR4
250 GB SSD or better
Linux

I/O
Front

2x USB 3.0 Host，
Calibration signal for passive probe: 1 kHz, 3 V Square
4x USB Host 3.1 Gen 1，2x 1000M LAN (VXI-11+SCPI， Telnet (5024) +SCPI，Socket (5025) +SCPI，
LXI，WebServer)

Side

1x DVI-D: up to 1920x1200 @ 60Hz，1x DP 1.2: up to 4096x2304 @ 60Hz，1x HDMI 1.4: up to4096x2160 @ 60Hz
Mic input, Audio Output
USB 2.0 Device（USBTMC）
External trigger in，EXT: ≤1.5 Vrms，EXT/5: ≤ 7.5Vrms,

Rear

Aux out: TRIG OUT(3.3 V LVCMOS)，PASS/FAIL OUT(3.3 V TTL),
10 MHz In，10 MHz Out
AWG

Display
Display Type
Resolution

15.6 HD TFT LCD with capacitive touch screen
1920×1080

Display Setting
Range
Multiple-window
Display Type
Persistence Time
Color Display
Language

WWW.SIGLENT.COM

8 x 10 grid
1x1, 2x1, 4x1, 1x2, 2x2, 4x2, 3x3Dot, Vector
OFF, 0.1 s, 0.2 s, 0.5 s, 1 s, 5 s, 10 s, 30 s, infinite
Normal, Color; Supports customer trace color
Simplified Chinese, Traditional Chinese, English, French, Japanese, German, Spanish, Russian,
Italian, Portuguese17

SDS7000A Series Digital Storage Oscilloscope
Built-in Help System

Simplified Chinese, English

Environmental
Temperature
Humidity
AltitudeElectromagnetic
Compatibility

SafetyRoHS

Operating: 0 °C ~ 50 °C
Non-operating: -30 °C ~ 60 °C
Operating: 5% ~ 90%RH, 30℃, degraded to 50%RH at 40 ℃
Non-operating: 5% ~ 95%
Operating: ≤ 3,048 m, 25 ℃
Non-operating: ≤12,192 m
Meets EMC directive (2014/30/EU), meets or exceeds IEC 61326-1:2012/EN61326-1:2013 (Basic)
CLASS A group 1
Conducted disturbance
CISPR 11/EN 55011
150 kHz-30 MHz
CLASS A group 1
Radiated disturbance
CISPR 11/EN 55011
30 MHz-1 GHz
Electrostatic discharge (ESD)
IEC 61000-4-2/EN 61000-4-2
4.0 kV (Contact),8.0 kV (Air)
10 V/m (80 MHz to 1 GHz);Radio-frequency
IEC 61000-4-3/EN 61000-4-3
3 V/m (1.4 GHz to 2 GHz);
electromagnetic field Immunity
1 V/m (2.0 GHz to 2.7GHz)
Electrical fast transients (EFT)IEC 61000-4-4/EN 61000-4-4
2kV (Input AC Power Ports)
1kV (Line to line)
SurgesIEC 61000-4-5/EN 61000-4-5
2kV (Line to ground)Radio-frequency continuous
IEC 61000-4-6/EN 61000-4-6
3 V, 0.15-80MHz
conducted Immunity
Voltage Dips:
0% UT during 1 cycle;
40% UT during 10/12 cycles;
Voltage dips and interruptions
IEC 61000-4-11/EN 61000-4-11
70% UT during 25/30 cycles
Voltage interruptions: 0% UTduring 250/300 cycles
UL 61010-1:2012/R: 2018-11; CAN/CSA-C22.2 No. 61010-1:2012/A1:2018-11.
UL 61010-2-030:2018; CAN/CSA-C22.2 No. 61010-2-030:2018.
EU 2015/863

Power Supply
Input Voltage & Frequency

100 ～ 240 Vrms 50/60HzPower consumption

400 W max., 210 W typical, 4 W typical in standby mode

Mechanical
Dimensions
Weight18

Width × Height × Depth
Without feet: 444.5mm × 334mm × 176.4mm
With feet: 444.5mm×367mm×176.4mm
Net Weight 10.6 kg,
Gross Weight 17.0 kg

WWW.SIGLENT.COM

SDS7000A Series Digital Storage Oscilloscope

Ordering Information
Model

Description

SDS7404A H12
SDS7304A H12

4 GHz, 20 GSa/s, 4-CH, 12-bit, standard 500 Mpts/ch memory depth, 15.6” capacitive touch screen
3 GHz, 20 GSa/s, 4-CH, 12-bit, standard 500 Mpts/ch memory depth, 15.6” capacitive touch screenStandard Accessories

Quantity

USB cable
Quick start
Passive probe (SP3150A)
Certificate of calibration
Wireless mouse
Power cordProtective Cover

11
1/channel
11
11

Optional Accessories

Part No.

16-channel logic probe
Power Analysis deskew fixture
USB 2.0 test fixture
Ethernet test fixture
Automotive Ethernet test fixture
STB3 demo signal source
USB-GPIB adapter
High-speed passive probe
High-speed active probe
High voltage probe
High-speed differential probe
High voltage differential probe

SPL2016
DF2001A
FX-USB2
FX-ETH
FX-AMETH
STB3
USB-GPIB
SP6150A
SAP1000, SAP2500
HPB4010
SAP2500D, SAP5000D
DPB1300/DPB4080/DPB5150/ DPB5150A/DPB5700/DPB5700A
CPL5100/CP4020/CP4050/CP4070/CP4070A
CP6030/CP6030A/CP6150/CP6500
SCP5030/SCP5030A/SCP5150/SCP5150A
SAP4000P
CASE-S2

Current probe
Power rail probe
Transit case

Options

Part No.

Waveform generator (software)Power Analysis (software)
Eye Diagram/Jitter Analysis (software)
I2S trigger & decode (software)
MIL-STD-1553B trigger & decode (software)
FlexRay trigger & decode (software)
CAN FD trigger & decode (software)
SENT trigger & decode (software)
Manchester decode (software)
ARINC429 trigger & decode (software)USB 2.0 decode (software)USB 2.0 compliance test (software)
100Base-TX compliance test (software)
1000Base-T compliance test (software)
100Base-T1 compliance test (software)
1000Base-T1 compliance test (software)

SDS7000A-FG
SDS7000A-PA
SDS7000A-EJ
SDS7000A-I2S
SDS7000A-1553B
SDS7000A-FlexRay
SDS7000A-CANFD
SDS7000A-SENT
SDS7000A-Manch
SDS7000A-ARINC
SDS7000A-USB2
SDS7000A-CT-USB2
SDS7000A-CT-100BASE-T
SDS7000A-CT-1000BASE-T
SDS7000A-CT-100BASE-T1
SDS7000A-CT-1000BASE-T1

1Gpts memory depth（software）

SDS7000A-1GPTS

OCXO timebase (Assembled and calibrated in factory only)

10M_OCXO_L

WWW.SIGLENT.COM

19

SDS7000A Series Digital Storage Oscilloscope

About SIGLENTSIGLENT is an international high-tech company, concentrating on R&D, sales,
production and services of electronic test & measurement instruments.

SIGLENT first began developing digital oscilloscopes independently in 2002.
After more than a decade of continuous development, SIGLENT has extended
its product line to include digital oscilloscopes, isolated handheld
oscilloscopes,

function/arbitrary

waveform

generators,

RF/MWsignalgenerators, spectrum analyzers, vector network analyzers, digital multimeters,
DC power supplies, electronic loads and other general purpose test
instrumentation. Since its first oscilloscope was launched in 2005, SIGLENT
has become the fastest growing manufacturer of digital oscilloscopes. We
firmly believe that today SIGLENT is the best value in electronic test &measurement.

Headquarters:
SIGLENT Technologies Co., Ltd
Add: Bldg No.4 & No.5, Antongda Industrial
Zone, 3rd Liuxian Road, Bao'an District,
Shenzhen, 518101, China
Tel: + 86 755 3688 7876
Fax: + 86 755 3359 1582
Email: sales@siglent.com
Website: int.siglent.com

North America:
SIGLENT Technologies America, Inc
6557 Cochran Rd Solon, Ohio 44139
Tel: 440-398-5800
Toll Free: 877-515-5551
Fax: 440-399-1211
Email: info@siglent.com

Follow us on
Facebook: SiglentTech

Website: www.siglentna.com

Europe:SIGLENT Technologies Germany GmbH
Add: Staetzlinger Str. 70
86165 Augsburg, Germany
Tel: +49(0)-821-666 0 111 0
Fax: +49(0)-821-666 0 111 22
Email: info-eu@siglent.com
Website: www.siglenteu.com
20

WWW.SIGLENT.COM