DIGITAL MULTIMETER

PC720M

APPLICATIONS AND FEATURES

This instrument is a portable multimeter designed for themeasurement of low-voltage circuit. This is used at small
communications equipments, home electric appliances,
voltage of lump line, various types of batteries, and circuit
analyses with additional functions.

●AC True RMS / 9999 count & 6000 count●0.06% best accuracy●Dual display capability with Backlight to simultaneously display
voltage/current/frequency and AC/DC components of voltage/current●High speed bar graph●Conductance measurement / Capacitance measurement●Temperature measurement(with K type thermocouple sensor : -50℃〜1000℃)●Frequency measurement(AC sine wave only)●Logic frequency measurement / Duty cycle measurement●MAX, MIN, AVE recording mode / Data hold / Range hold
●Relative value●Auto power save(30min after the last operation)(cancelable)●Backlight●Protective holster with wall hanger and lead holder / Tilt stand●PC Link system with optical Link USB cable (KB-USB7) and software(PC Link7)*KB-USB7 and PC Link7 are optional accessories

Data Logging Mode
●Built-in memory:87,328 data points(single display)/43,664 data point(dual display)●Selection of measurement interval
0.05s/0.1s/0.5s/1s/2s/3s/4s/5s/10s/15s/30s/60s/120s/180s/300s/600s●Auto-standby mode when a sampling speed of 30s or longer is selected●Export logged data to PC*Optional accessories, KB-USB7 and PC Link7 are necessary

SPECIFICATIONS
DCV
ACV
DCA
ACA
Resistance
Capacitance
Temperature
Frequency
Logic frequency
Duty cycle
Continuity
Diode test

Measuring range
Best accuracy
Resolution
60m/600m/9.999/99.99/999.9V±(0.06%+2)
0.01mV
60m/600m/9.999/99.99/999.9V±(0.5%+3)
0.01mV
600μ/6000μ/60m/600m/6/10A±(0.2%+4)0.1μA
600μ/6000μ/60m/600m/6/10A±(0.6%+3)
0.1μA
600/6k/60k/600k/6M/60MΩ/99.99nS *1±(0.1%+3)0.1Ω
±(0.8%+3)*2
0.01nF60n/600n/6μ/60μ/600μ/6m/25mF
1℃-50℃〜1000℃(thermocouple K type)±(0.3%+2)
0.01Hz
15Hz〜50kHz±(0.04%+4)0.001Hz
5Hz〜1MHz±(0.03%+4)
0.01%
0%〜100%±(3d/kHz+2)
Buzzer sounds at between 20Ω and 300Ω Open voltage: approx. 1.2V
Open voltage: approx. 3.5V

Input impedance
10MΩ

Battery low warning
Sampling rate
Power consumption
Bandwidth*1 nS(Conductance): High-value resistance of Giga-Ohms for leakage measurements. Conductance is the inverse of Resistance, that is S=1/Ω or nS=1/GΩ
*2 Accuracy of film capacitor or equivalent with low leakage.

A battery for monitoring has been installed prior to shipment from the factory. It may be discharged before the expirationof the described battery life. This battery is used to check the functions and performance of the product.Specifications
and external appearance of the product described above may be revised for modification without prior notice.

Distributed by

Dempa Bldg, 4-4 Sotokanda 2-Chome, Chiyoda-Ku, Tokyo 101-0021 Japan
Tel:+81-3-3251-0941 Fax:+81-3-3256-9740

www.sanwa-meter.co.jp

Operation method
AC measuring method
Display
Range selection
Overload
Polarity

EN61010-1 CAT.II
EN61010-1 CAT.III
Fuse / Battery
Size / Mass
Standard accessories
includedΔ-ΣTrue RMS
6000 count and 9999 countAuto and manual ranges"O.L" is displayed
Automatic selection(only " - " is displayed whennegative voltage is inputted.)
Battery() mark lights when the internal battery's
power is below approx. 7V
Numeric part : 5 times/sec
Bar graph part : 60 times/secApprox. 48mW Approx. 0.45mW(Auto power save)
V : 40Hz〜3kHz, 3kHz〜20kHz(below 99.99V)
A : 40Hz〜1kHz
1000V
600V
11A/1000V IR20kA Φ10x38mm
6LR61(9V) x 1
0.4A/1000V IR30kA Φ6.3x32mm
H184 x W86 x D52mm / 430g
Test Lead(TL-23a), Holster(H-700),
Thermocouple K type (K-250PC), Instruction Manual