SDS2000X HD
Digital Storage
Oscilloscope

Data Sheet
2022.02

EN02A

Key Features
12-bit High Resolution

SDS2354X HD
SDS2204X HD

12-bit Analog-Digital Convertors with sample rate up
to 2 GSa/s

Front ends with 70 μVrms noise floor @ 500 MHz
bandwidth and 0.5% DC gain accuracy

4 analog channels, up to 350 MHz bandwidth (upgradable
to 500 MHz)SPO technology

Product Overview
SIGLENT’s SDS2000X HD series high resolution digital
storage oscilloscopes are based on 2 GSa/s, 12-bit AnalogDigital Converters and front ends with excellent noise floor
performance. They are available in bandwidths of 350 MHz and
200 MHz, have maximum record length of 200 Mpts/ch, anddisplay 4 analog channels + 16 digital channels mixed signal
analysis ability.
The SDS2000X HD series employs Siglent’s SPO technology
with a maximum waveform capture rate of up to 100,000 wfm/s (normal mode, up to 500,000 wfm/s in Sequence mode), 256level intensity grading display function plus a color temperature
display mode. It also employs an innovative digital trigger
system with high sensitivity and low jitter. The trigger system
supports multiple powerful triggering modes including serial
bus triggering. Tools such as History waveform recording,Search and Navigate functions, Mask Test, Bode Plot, Power
Analysis and Histogram allow for extended waveform records
to be captured, stored, and analyzed. An impressive array ofmeasurement and math capabilities, options for a 25 MHz
arbitrary waveform generator, as well as serial decoding arealso features of the SDS2000X HD.
The large 10.1’’ display capacitive touch screen supports multitouch gestures, with the addition of user-friendly UI design, cangreatly improve the operation efficiency. It also supports mouse
control, and remote web control over LAN.



Waveform capture rate up to 100,000 wfm/s (normalmode), and 500,000 wfm/s (sequence mode)Supports 256-level intensity grading and color
temperature display modesUp to 200 Mpts/ch record length

Digital trigger system

Intelligent trigger: Edge, Slope, Pulse, Window, Runt,
Interval, Dropout, Pattern, Qualified, Nth edge,
Setup/hold, Delay and Video (HDTV supported). Zone
Trigger simplifies advanced triggering
Serial bus triggering and decoder, supports protocols I2C,
SPI, UART, CAN, LIN, CAN FD, FlexRay, I2S, MIL-STD1553B, SENT, Manchester and ARINC429
Segmented acquisition (Sequence) mode, dividing the
maximum record length into multiple segments (up to
80,000), according to trigger conditions set by the user,
with a very small dead time between segments to capture
the qualifying event
History waveform record (History) function, the maximum
recorded waveform length is 80,000 frames
Automatic measurements on 50+ parameters, supports
statistics
withhistogram, track, trend, Gating
measurement, and measurements on Math, History andRef
4 Math traces (2 Mpts FFT, addition, subtraction,
multiplication, division, integration, differential, squareroot, etc.), supports formula editor
Abundant data analysis functions such as Search,
Navigate, Digital Voltmeter, Counter, Waveform
Histogram, Bode plot and Power Analysis
High Speed hardware-based Average, ERES; High
Speed hardware-based Mask Test function, with Mask
Editor tool for creating user-defined masks
16 digital channels (optional)
Built-in 25 MHz waveform generator
Large 10.1” TFT-LCD display with 1024 * 600 resolution;
Capacitive touch screen supports multi-touch gestures
Interfaces include: USB Hosts, USB Device (USBTMC),
LAN (LXI/VXI-11/Telnet/Socket), Pass/Fail, Trigger Out
Built-in web server supports remote control over the LAN
port using a web browser. Supports SCPI remote control
commands. Supports external mouse and keyboard

WWW.SIGLENT.COM

1

SDS2000X HD Series Digital Storage Oscilloscope

Models and Key Specifications
Model

SDS2354X HD

Analog channels

4 + EXT
350 MHz(Upgradable to 500 MHz max.)
12-bitBandwidth
Vertical resolution
Sample rate
(Max.)
Memory depth(Max.)SDS2204X HD
200 MHz

2 GSa/s (interleaving mode）, 1 GSa/s (non-interleaving mode)
200 Mpts/ch (interleaving mode)，100 Mpts/ch (non-interleaving mode)

Waveform capture
rate (Max.)

Normal mode：100,000 wfm/s;

Trigger type

Edge, Slope, Pulse width, Window, Runt, Interval, Dropout, Pattern, Video, Qualified, Nth edge,
Setup/hold, Delay, Serial

Serial trigger anddecodeStandard: I2C, SPI, UART, CAN, LIN
Optional: CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT, Manchester (decode only), ARINC429 (decode
only)
50+ parameters, statistics, histogram, trend, and track supported

Measurement

Sequence mode：500,000 wfm/s

4 traces
MathData analysis
Digital channel(optional)
Waveform
generator(optional)
I/O
Probe (Standard)
Display2

2 Mpts FFT, +, -, x, ÷, ∫dt, d/dt, √, Identity, Negation, Absolute, Sign, ex, 10x, ln, lg, Interpolation, MaxHold,
MinHold, ERES, Average, Filter. Supports formula editor
Search, Navigate, History, Mask Test, Digital Voltmeter, Counter, Waveform Histogram, Bode plot, andPower Analysis
16-channel; maximum sample rate up to 500 MSa/s; record length up to 50 Mpts
Single-channel built-in waveform generator, frequency up to 25 MHz, 125 MSa/s sample rate, 16 kpts
waveform memory
USB 2.0 Host x3, USB 2.0 Device, 10 M / 100 M LAN, External trigger, Auxiliary output (TRIG OUT,
PASS/FAIL)
One 500 MHz passive probe supplied for each channel
10.1 TFT-LCD with capacitive touch screen (1024*600)

WWW.SIGLENT.COM

SDS2000X HD Series Digital Storage Oscilloscope

Functions & Characteristics
12-bit High Resolution

12-bit8-bit

12-bit resolution shows you more details and less noise on the waveform.

Low noise floor: Only 70 µVrms at 500 MHz bandwidth

0.5% DC gain accuracy

Excellent User Interface and User Experience

WWW.SIGLENT.COM



10.1” display with 1024x600 resolution

Capacitive touch screen, supporting multi-touchgestures, can move or scale the waveform traces
quickly by finger-touch movements, which greatly
improves the operation efficiency



Built-in WebServer supports remote control on a
web page over LAN

Supports external mouse and keyboard

3

SDS2000X HD Series Digital Storage OscilloscopeHigh Waveform Update Rate

Deep Record LengthWith a waveform update rate of up to 100,000 wfm/s, theoscilloscope can easily capture unusual or low-probability
events. In Sequence mode, the waveform capture rate canreach 500,000 wfm/s

Using hardware-based Zoom technique and record length of up
to 200 Mpts, users can select a slower timebase withoutcompromising the sample rate, and then quickly zoom in tofocus on the area of interest

Multiple Trigger Functions

Trigger Zone

Edge, Slope, Pulse, Video, Windows, Runt, Interval, Dropout,
Pattern, Qualified, Nth edge, Setup/hold, Delay, and serial
trigger

Trigger Zone is available for advanced triggering

Advanced Math Function

In addition to the traditional (+, -, X, /) operations, FFT,
integration, differential, square root, and more are supported.
Formula Editor is available for more complex operations. 4 mathtraces are available.

4

Hardware-accelerated FFT supports up to 2 Mpts operation.
This provides high-frequency resolution with a fast refresh rate.
The FFT function also supports a variety of window functions sothat it can adapt to different spectrum measurement needs.
Three modes (Normal, Average, and Max hold) can satisfy
different requirements for observing the power spectrum. Autopeak detection and markers are supported.

WWW.SIGLENT.COM

SDS2000X HD Series Digital Storage Oscilloscope

Measurements of a Variety of Parameters

Parameter measurements include 4 categories: horizontal,
vertical, miscellaneous, and CH delay providing a total of 50+
different types of measurements. Measurements can be
performed within a specified gate period. Measurements on
Math, Reference, and History frames are supported

Parameter Statistics Function

Statistics show the current value, maximum value, minimum
value, standard deviation, and mean value of up to 12
parameters simultaneously. A histogram is available to show the
probability distribution of a parameter. Trend and Track areavailable to show the parameter value vs. time.
For horizontal parameters such as period, all results are
extracted from a frame, instead of just calculating the first one.
This accelerates statistics on horizontal measurements much
more and enables distribution observation in a frame using
Histogram and Track

History Mode

Sequence Mode

History function can record up to 80,000 frames of waveforms.
The recording is executed automatically so that the customer
can playback the history waveforms at any time to observe
unusual events and quickly locate the area of interest using the
cursors or measurements. The failed frames of the Mask Test
can be stored as history

Segmented memory collection will store the waveform into
multiple memory segments (up to 80,000) and each segment
will store a triggered waveform as well the dead time
information. The interval between segments can be as small as
2 μs. All of the segments can be played back using the History
function

WWW.SIGLENT.COM

5

SDS2000X HD Series Digital Storage Oscilloscope

Search and Navigate

Serial Bus Decode

The oscilloscope can search events specified by the user in aframe. Events flagged by the Search can be recalled
automatically using Navigate. It can also navigate by time (delay
position) and history frames

Display the decoded characters through the events list. Bus
protocol information can be quickly and intuitively displayed intabular form. I2C, SPI, UART, CAN, LIN, CAN FD, FlexRay, I2S,
MIL-STD-1553B, SENT, Manchester and ARINC429 aresupported

Hardware-based High-Speed Mask Test Function

The oscilloscope utilizes a hardware-based Mask Test function,
performing up to 14,000 Pass / Fail decisions each second. It iseasy to generate user-defined test templates to provide trace
mask comparisons, making it suitable for long-term signal
monitoring or automated production line testing

6

Built-in Mask Editor application helps to create custom masks

WWW.SIGLENT.COM

SDS2000X HD Series Digital Storage Oscilloscope

Bode PlotPower Analysis (Optional)

The oscilloscope can control the built-in waveform generator ora stand-alone SIGLENT generator, to scan the amplitude andphase-frequency response of the DUT, and display the data asa Bode Plot. This makes it possible to replace expensive
network analyzers in some applications

The Power Analysis option provides a full suite of power
measurements and analysis, which greatly improve themeasurement efficiency in switching power supplies and powerdevices design

Digital Channels / MSO (Optional)

Built-in 25 MHz Waveform Generator (Optional)

Four analog channels plus 16 digital channels enable users toacquire and trigger the waveforms then analyze the pattern,simultaneously with one instrument

The built-in waveform generator can output waveforms with up
to 25 MHz frequency and ±3 V amplitude. Six basic waveforms
together with multiple types of predefined waveforms and asuser-defined arbitrary waveforms are supported.

Complete Connectivity

USB Host 2.0 x3, USB Device 2.0 (USBTMC) , LAN (VXI-11/Telnet/Socket), Auxiliary output (Pass/Fail, Trigger Out), etc.

WWW.SIGLENT.COM

7

SDS2000X HD Series Digital Storage Oscilloscope

Specifications
All specifications are not guaranteed unless the following conditions are met:
• The oscilloscope calibration period is current• The oscilloscope has been working continuously for at least 30 minutes at the specified temperature (18℃ ~ 28℃)

Acquire (analog)
Sample rate

2 GSa/s (interleaving mode* 1），1 GSa/s (non-interleaving mode* 2 )Memory depth

200 Mpts/ch (interleaving mode)， 100 Mpts/ch (non-interleaving mode) * 3

Memorymanagement
mode

Auto, Fixed sample rate, Fixed memory length

Waveform update
rate

Normal mode：up to 100,000 wfm/s

Intensity grading
Peak detect
Average
ERES
Sequence
History
Interpolation

256-level
1 ns
4, 16, 32, 64, 128, 256, 512, 1024
Enhanced bit: 0.5, 1, 1.5, 2, 2.5, 3 bits
Up to 80,000 segments, interval between triggers = 2 μs min.
Up to 80,000 frames
sinx/x, x

Sequence mode：up to 500,000 wfm/s* 1：Interleaving mode: only one of C1/C2 and/or only one of C3/C4 activated* 2：Non-interleaving mode: both C1/C2 and/or both C3/C4 activated* 3：In Average and ERES modes, the memory depth is 20 Mpts/ch (interleaving mode), 10 Mpts/ch (non-interleaving mode)

Vertical(analog)
ChannelBandwidth(-3 dB) @ 50 Ω
Flatness@50 Ω
Rise time@50 Ω(typical)Resolution
ENOB *2 (typical)
Noise floor*3 (rms,@50 Ω, typical,1mV/div)
Range
Vertical scale(probe 1X)
DC gain accuracy
Offset accuracy

Offset range(probe 1X)

SDS2354X HD
4 + EXT
350 MHz(Upgradable to 500 MHz max.) *1
10 kHz ~ BW/10: ±0.5 dB
BW/10 ~ BW/3: ±0.8 dB
BW/3 ~ BW2/3: +1.0 dB, -1.2 dB
BW2/3 ~ BW: +2.0 dB, -2.5 dB

SDS2204X HD
200 MHz1 ns

1.7 ns12-bit8.4-bit

8.6-bit

65 μV

55 μV

8 divisions
1 MΩ: 0.5 mV/div – 10 V/div
50 Ω: 0.5 mV/div – 1 V/div
0.5 mV/div ~ 4.95 mV/div: ±1.5 %;5 mV/div ~ 10 V/div: ±1.0 % maximum, ±0.5 % typical;± (0.5% of the offset setting + 0.5% of full scale + 1 mV)
1 MΩ:

50 Ω:

0.5 mV/div ~ 5 mV/div: ±1.6 V;

0.5 mV/div ~ 5 mV/div: ±1.6V;

5.1 mV/div ~ 10 mV/div: ±4 V;

5.1 mV/div ~ 10 mV/div: ±4 V;

10.2 mV/div ~ 20 mV/div:±8 V;

10.2 mV/div ~ 20 mV/div:±8 V;

20.5 mV/div ~ 100 mV/div: ±16 V；

20.5 mV/div ~ 1 V/div: ±10 V

102 mV/div ~ 200 mV/div: ±80 V;
205 mV/div ~ 1 V/div: ±160 V;
1.02 V/div ~ 10 V/div: ±400 V
Bandwidth limit
Low frequency
8

Hardware Bandwidth limit: 20 MHz, 200 MHz
5 Hz (typical)
WWW.SIGLENT.COM

SDS2000X HD Series Digital Storage Oscilloscope
response (ACcoupling -3 dB)
Overshoot (100mV/div, 150 psedge @50 Ω,
typical)Coupling
Impedance

12%

3%

DC, AC, GND
1 MΩ: (1 MΩ±2%) || (16 pF±2 pF)
50 Ω: 50 Ω±1%Max. input
voltage

1 MΩ ≤ 400 Vpk (DC + AC), DC~10 kHz

SFDR≥ 45dBc

CH to CH
Isolation (@50Ω)> 60 dBc, < 500 MHz> 70 dBc, < 350 MHz
1X, 10X, 100X, custom

Probe Attenuation

50 Ω ≤ 5V rms* 1：In interleaving mode bandwidth is 500 MHz; in non-interleaving mode bandwidth is 350 MHz* 2：99.99 MHz input，-0.5 dBFS，20 mV/div，50 Ω input impedance* 3：Use the “Stdev” measurement @ 2 GSa/s, 10 Mpts/ch

Horizontal
Time scale
Range
Display mode
Roll mode
Skew (C1~C4)
Time base
Accuracy

1 ns/div – 1000 s/div
10 divisions
Y-T, X-Y, Roll
≥ 50 ms/div< 100 ps
±2 ppm initial (0~50℃); ±0.5 ppm 1st year aging; ±3 ppm 20-year aging

TriggerMode

Auto, Normal, Single
Internal: ±4.1 div from the center of the screen

Level

EXT: ± 0.61 V
EXT/5: ± 3.05 V

Ext Trigger input
voltage

1 MΩ ≤ 42 Vpk

Hold off range

By time: 8 ns ~ 30 s (8 ns step)
By event: 1 ~ 108
C1~C4
DC: Passes all components of the signal
AC: Blocks DC components and attenuates signals below 7 Hz
LFRJ: Attenuates the frequency components below 1.17 MHz
HFRJ: Attenuates the frequency components above 660 kHz
Noise RJ: Increases the trigger hysteresis

Coupling

50 Ω ≤ 5V rms

EXT
DC: Passes all components of the signal
AC: Blocks DC components and attenuates signals below 18 Hz
LFRJ: Attenuates the frequency components below 7.5 kHz
HFRJ: Attenuates the frequency components above 250 kHz
Accuracy (typical)

C1～C4: ±0.2 div
EXT: ±0.3 div
C1 ～ C4:

Sensitivity

EXT:

EXT/5:
JitterWWW.SIGLENT.COM>10 mV/div:
5 mV/div~10 mV/div:≤ 2 mV/div:

Noise RJ = OFF±0.26 div
±0.26 div±0.5 div

Noise RJ = ON
±0.33 div
±0.33 div±0.5 div

200 mVpp, DC ～ 200 MHz
500 mVpp, 200 MHz ～ bandwidth（300 MHz）
1 Vpp, DC ～ 200 MHz
1.5 Vpp, 200 MHz ～ bandwidth（300 MHz）

C1～C4: < 10 ps rms (typical) ，≥ 6 div Vpp sine，2.5 mV/div ~ 10 V/div
9

SDS2000X HD Series Digital Storage Oscilloscope
EXT: < 200 ps rms
Displacement

Pre-Trigger: 0 ~ 100% memory
Delay-Trigger: 0 ~ 10,000 div
Up to 2 zones

Zone
Edge Trigger
SourceSlope
Slope Trigger
SourceSlope
Limit range

Source: C1~C4
Property: Intersect, Not Intersect
C1~C4/EXT/(EXT/5)/AC Line/D0~D15
Rising, Falling, Rising & Falling
C1~C4
Rising, Falling<, >, in range, out of range

Time range

2 ns ～ 20 s, Resolution = 1 ns

Pulse Width Trigger
SourcePolarity
Limit range

C1~C4/D0~D15+width, -width<, >, in range, out of range

Time range

2 ns ～ 20 s, Resolution = 1 ns

Video Trigger
SourceStandard
Synchronization
Trigger Condition
Window Trigger
SourceWindow typeInterval Trigger
SourceSlope
Limit range

C1~C4
NTSC, PAL, 720p/50, 720p/60, 1080p/50, 1080p/60, 1080i/50, 1080i/60, Custom
Any, SelectLine, Field
C1~C4
Absolute, Relative
C1~C4/D0~D15
Rising, Falling<, >, in range, out of range

Time range

2 ns ～ 20 s, Resolution = 1 ns

Dropout Trigger
SourceTimeout type
Slope

C1~C4/D0~D15
Edge, State
Rising, Falling

Time range

2 ns ～ 20 s, Resolution = 1 ns

Runt Trigger
SourcePolarity
Limit range

C1~C4
Positive, Negative<, >, in range, out of range

Time range

2 ns ～ 20 s, Resolution = 1 ns

Pattern Trigger
SourcePattern Setting
Logic
Limit range

C1~C4/D0~D15
Don’t Care, Low, HighAND, OR, NAND, NOR<, >, in range, out of range

Time range

2 ns ～ 20 s, Resolution = 1 ns

Qualified Trigger
Type
Qualified Source
Edge Trigger Source
Nth Edge Trigger
SourceSlope
Idle time

State, State with Delay, Edge, Edge with Delay
C1~C4/D0~D15
C1~C4/D0~D15
C1~C4/D0~D15
Rising, Falling
8 ns ～ 20 s, Resolution = 1 ns

Edge Number

1 ～ 65535

Delay Trigger
Source A
Source B

C1~C4/D0~D15
C1~C4/D0~D15

10WWW.SIGLENT.COM

SDS2000X HD Series Digital Storage Oscilloscope
Slope
Limit range

Rising, Falling<, >, in range, out of range

Time range

2 ns ～ 20 s, Resolution = 1 ns

Serial Trigger
Source

I2 C
SPI
UART
CAN
LIN

C1~C4/D0~D15
Standard: I2C, SPI, UART, CAN, LIN
Optional: CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT
Type: Start, Stop, Restart, No Ack, EEPROM, Address & Data, Data Length
Type: Data
Type: Start, Stop, Data, Parity Error
Type: All, Remote, ID, ID+Data, Error
Type: Break, Frame ID, ID+Data, Error

CAN FD（Optional）

Type: Start, Remote, ID, ID+Data, Error

FlexRay（Optional）

Type: TSS, Frame, Symbol, Errors

I2S（Optional）

Type: Data, Mute, Clip, Glitch, Rising Edge, Falling Edge

MIL-STD-1553B（Optional）

Type：Transfer, Word, Error, Timing

SENT（Optional）

Type: Start, Slow channel, Fast channel, Error

ProtocolSerial Decoder
Decoders

2

Threshold-4.1 ～ 4.1 div

List

1 ～ 7 lines

Decoder type
I2 C
SourceSignalAddress
SPI
SourceSignalEdge SelectChip selectBit Order
UART
SourceSignalData Width
Parity Check
Stop Bit
Idle Level
Bit Order
CAN
SourceLIN
LIN Version
SourceBaud Rate

Full duplex
C1~C4/D0~D15
SCL, SDA
7-bit, 10-bit
C1~C4/D0~D15
CLK, MISO, MOSI, CS
Rising, Falling
Active high, Active low, Clock timeout
LSB, MSB
C1~C4/D0~D15
RX, TX
5-bit, 6-bit, 7-bit, 8-bit
None, Odd, Even, Mark, Space
1-bit, 1.5-bit, 2-bit
Low, High
LSB, MSB
C1~C4/D0~D15
Ver 1.3, Ver 2.0
C1~C4/D0~D15
600 bps, 1200 bps, 2400 bps, 4800 bps, 9600 bps, 19200 bps, Custom

CAN FD（Optional）
SourceNominal Baud Rate
Data Baud Rate

C1~C4/D0~D15
10 kbps, 25 kbps, 50 kbps, 100 kbps, 250 kbps, 1 Mbps, Custom
500 kbps, 1 Mbps, 2 Mbps, 5 Mbps, 8 Mbps, 10 Mbps, Custom

FlexRay（Optional）
SourceBaud Rate

C1~C4/D0~D15
2.5 Mbps, 5 Mbps, 10 Mbps, Custom

I2S（Optional）
SourceSignalAudio Variant
Start Bits
WWW.SIGLENT.COM

C1~C4/D0~D15
BCLK, WS, DATA
Audio-I2S, Audio-LJ, Audio-RJ
0~31
11

SDS2000X HD Series Digital Storage Oscilloscope
Data Bits
1~32
MIL-STD-1553B（Optional）
SourceC1~C4

SENT（Optional）
SourceC1~C4/D0~D15

Manchester（Optional）
SourceBaud Rate

C1~C4
500 bps~5 Mbps

ARINC429（Optional）
SourceBaud Rate

C1~C4
12.5 kbps~100 kbps, tolerance 1%~20%

Word format

L/SDI/D/SSM，L/D/SSM，L/D

Measurement
Automatic Measurement
SourceC1~C4, D0~D15, Z1~Z4, F1~F4, M1~M4, Ref, History
Mode
Simple, Advanced
ScreenRange
Gated: inside screen, definable with separate Gate cursors
Custom Threshold
Upper, Middle, Lower
No. of Measurements
Display 12 measurements at the same time (Display mode = M2)
Max, Min, Pk-Pk, Top, Base, Amplitude, Mean, Cycle Mean, Stdev, Cycle Stdev, RMS, Cycle RMS,
Vertical Parameters
Median, Cycle Median, FOV, FPRE, ROV, RPRE, Level@Trigger
Period, Frequency, Time@max, Time@min, +Width, -Width, 10-90%Rise time, 90-10%Fall time, Rise
Horizontal Parameters
time, Fall time, +Burst Width, -Burst Width, +Duty Cycle, -Duty Cycle, Delay, Time@Middle, CycleCycle jitter+Area@DC, -Area@DC, Area@DC, Absolute Area@DC, +Area@AC, -Area@AC, Area@AC,
Miscellaneous
Absolute Area@AC, Cycles, Rising Edges, Falling Edges, Edges, Positive pulses, Negative pulses,
Parameters
Positive Slope, Negative Slope
Phase, FRFR, FRFF, FFFR, FFFF, FRLR, FRLF, FFLR, FFLF, Skew, Tsu@R, Tsu@F, Th@R,
Delay Parameters
Th@F
Statistics
Current, Mean, Min, Max, Sdev, Count, Histogram, Trend, Track
Statistics Count
Unlimited, 1~1024
Statistics Count in one
Up to 25,000
frame
Cursors
SourceC1~C4, D0~D15, F1~F4, M1~M4, Ref, Histogram
Manual : Time X1, X2, (X1-X2), (1/ΔT); Vertical Y1, Y2, (Y1-Y2)
Type
Track: Time X1, X2, (X1-X2)
Measure: indicates the measurement on specific parameter

Math
Trace
SourceOperation

FFT

F1~F4
C1~C4, M1~M4, F1~F4
FFT, +, -, x, ÷, ∫dt, d/dt, √, Identity, Negation, |x|, Sign, ex, 10x, ln, lg, Interpolation, Max hold, Min
hold, ERES, Average, Filter, Formula Editor
Length: 2 Mpts, 1 Mpts, 512 kpts, 256 kpts, 128 kpts, 64 kpts, 32 kpts, 16 kpts, 8 kpts, 4 kpts, 2 kpts
Window: Rectangular, Blackman, Hanning, Hamming, Flattop
Display: Full Screen, Split, Exclusive
Mode: Normal, Max hold, Average
Tools: Peaks, Markers

Analysis
SearchSourceMode
Copy setting
Navigate
Type
Mask Test

C1~C4, History
Edge, Slope, Pulse, Interval, Runt
Copy from trigger, Copy to trigger
Search event, Time, History frame

SourceC1~C4，Z1~Z4

Mask creating
Mask test speed

Auto (Create mask), Customized (Mask Editor)
Up to 14,000 frames/s

12

WWW.SIGLENT.COM

SDS2000X HD Series Digital Storage Oscilloscope
DVM
Source
Mode
Plot
Gate
Bode Plot
SourceC1~C4
DC mean, DC RMS, AC RMS, Peak-peak, Amplitude
Bar, Histogram, Trend
20 ms

C1~C4
Built-in waveform generator
Supported signal
SAG1021I (Connection: USB),
sources
SDG series waveform generators (Connection: USB, LAN)
Sweep type
Simple, Vari-level
Mode: Linear, Logarithmic
Frequency
Range: 10 Hz ~ 120 MHz
MeasureUpper cutoff frequency, Lower cutoff frequency, Bandwidth, Gain margin, Phase margin
Power Analysis (optional)
Power quality, Current Harmonics, Inrush current, Switching loss, Slew rate, Modulation, Output
Measureripple, Turn on/turn off, Transient response, PSRR, Efficiency, SOA
Histogram
SourceC1~C4
Type
Horizontal, Vertical, BothCounter
SourceC1~C4Frequency resolution
7 digits
Totalizer
Counter on edges, supports Gate and Trigger

Digital Channels（optional）
Max. Sampling Rate
Memory Depth
Min. Detectable Pulse
Width
Level Group
Level Range
Logic Type
Skew

500 MSa/s
50 Mpts/ch
3.3 ns
D0~D7, D8~D15-10 V~10 V
TTL, CMOS, LVCMOS3.3, LVCMOS2.5, Custom
D0~D15: ±1 sampling interval
Digital to Analog: ± (1 sampling interval +1 ns)

Waveform Generator (optional)
ChannelsMax. Output Frequency
Sampling Rate
Frequency Resolution
Frequency Accuracy
Vertical Resolution
Amplitude Range
Waveforms
Output ImpedanceProtection
Sine
Frequency
Offset accuracy (10kHz)
Amplitude flatness
SFDR
Harmonic distortion1
25 MHz
125 MSa/s
1 μHz±50 ppm
14 bit-1.5 V ~ +1.5 V (into 50 Ω)-3 V ~ +3 V (into High-Z)
Sine, Square, Ramp, Pulse, DC, Noise, 45 Arbitrary
50 Ω ± 2%
Over voltage protection, Current limit
1 μHz ～ 25 MHz±(1%*offset setting value +3 mVpp)±0.3 dB, compare to 10 kHz, 2.5 Vpp into 50 Ω
DC ~ 1 MHz-60 dBc
1 MHz ~ 5 MHz -55 dBc
5 MHz ~ 25 MHz -50 dBc
DC ~ 5 MHz-50 dBc
5 MHz ~ 25 MHz-45 dBc

Square/Pulse
Frequency

1 μHz ～ 10 MHz

Duty cycle

1% ～ 99%

Edge< 24 ns (10% ～ 90%)

Overshoot< 3% (typical，1 kHz, 1 Vpp)

WWW.SIGLENT.COM

13

SDS2000X HD Series Digital Storage Oscilloscope
Pulse width> 50 ns
Jitter (cycle-cycle)< 500 ps + 10 ppm
Ramp
Frequency
1 μHz ～ 300 kHz
Linearity
Channels< 0.1% of Pk-Pk (typical, 1 kHz, 1 Vpp, 50% symmetry)
0% ～ 100%

DC
Offset range
Accuracy
Noise
Bandwidth (-3 dB)
Arb
Frequency
Waveform memory
Sample rate
Wave import±1.5 V (into 50 Ω)±3 V (into Hi-Z)±(|setting value|*1% + 3 mV)>25 MHz
1 μHz ～ 5 MHz
16 kpts
125 MSa/s
From EasyWaveX, from U-disk, directly from waveform data of analog channels

I/O
Front

Rear

USB 2.0 Host x2，
Calibration Signal: 1 kHz,3 V SquareUSB 2.0 Host,
USB 2.0 Device,
LAN: 10/100MbaseT (RJ45),
External Trigger，EXT: ≤1.5 Vrms，EXT/5: ≤ 7.5Vrms,
Auxiliary Output: TRIG OUT(3.3 V LVCMOS), PASS/FAIL OUT(3.3 V TTL)

Display
Display Type
Resolution
Contrast (typical)
Backlight (typical)

10.1 TFT LCD with capacitive touch screen
1024×600
500:1
500 nit

Display Setting
Range
Display Type
Persistence Time
Color Display
Language
Built-in Help System

8 x 10 gridDot, Vector
OFF, 0.1 s, 0.2 s, 0.5 s, 1 s, 5 s, 10 s, 30 s, infinite
Normal, Color;
Supports customer trace color
Simplified Chinese, Traditional Chinese, English, French, Japanese, German, Spanish, Russian,
Italian, Portuguese
Simplified Chinese, English

Environmental
Temperature
Humidity
AltitudeElectromagnetic
Compatibility

14

Operating: 0 °C ~ 50 °C
Non-operating: -30 °C ~ 70 °C
Operating: 5% ~ 90%RH, 30℃, degraded to 50%RH at 50 ℃
Non-operating: 5% ~ 95%
Operating: ≤ 3,000 m, 25 ℃
Non-operating: ≤15,000 m
Meets EMC directive (2014/30/EU), meets or exceeds IEC 61326-1:2012/EN61326-1:2013 (Basic)
CLASS A group 1
Conducted disturbance
CISPR 11/EN 55011
150 kHz-30 MHz
CLASS A group 1
Radiated disturbance
CISPR 11/EN 55011
30 MHz-1 GHz
Electrostatic discharge (ESD)
IEC 61000-4-2/EN 61000-4-2
4.0 kV (Contact),8.0 kV (Air)
10 V/m (80 MHz to 1 GHz);Radio-frequency
IEC 61000-4-3/EN 61000-4-3
3 V/m (1.4 GHz to 2 GHz);
electromagnetic field Immunity
1 V/m (2.0 GHz to 2.7 GHz)Electrical fast transients (EFT)IEC 61000-4-4/EN 61000-4-4
2kV (Input AC Power Ports)
1kV (Line to line)
SurgesIEC 61000-4-5/EN 61000-4-5
2kV (Line to ground)Radio-frequency continuous
IEC 61000-4-6/EN 61000-4-6
3 V, 0.15-80 MHz
conducted Immunity
Voltage dips and interruptions
IEC 61000-4-11/EN 61000-4-11 Voltage Dips:
WWW.SIGLENT.COM

Safety
RoHS

SDS2000X HD Series Digital Storage Oscilloscope
0% UT during 1 cycle;
40% UT during 10/12 cycles;
70% UT during 25/30 cycles
Voltage interruptions: 0% UTduring 250/300 cycles
UL 61010-1:2012/R: 2018-11; CAN/CSA-C22.2 No. 61010-1:2012/A1:2018-11.
UL 61010-2-030:2018; CAN/CSA-C22.2 No. 61010-2-030:2018.
EU 2015/863

Power Supply
Input Voltage & Frequency

100 ～ 240 Vrms 50/60 Hz

Power consumption

120 W max., 70 W typical, 4 W typical in standby mode

Mechanical
Length × Height × Width = 317.2 mm × 236.0 mm × 149.0 mm
Dimensions
WeightWWW.SIGLENT.COM(including knobs and supporting legs)
Net Weight 4.1 kg, Gross Weight 5.6 kg

15

SDS2000X HD Series Digital Storage Oscilloscope

Ordering Information
Model

Description

SDS2354X HD
SDS2204X HD

12-bit, 350 MHz, 2 GSa/s, 4-CH, 200 Mpts/ch memory depth, 10.1” capacitive touch screen
12-bit, 200 MHz, 2 GSa/s, 4-CH, 200 Mpts/ch memory depth, 10.1” capacitive touch screenStandard Accessories

Quantity

USB cable
Quick start
Passive probe ( 500 MHz)
Certificate of calibration
Wireless mouse
Power cord

111/channel
11
1

Optional Accessories

Part No.

16-channel logic probe
Power Analysis deskew fixture
STB3 demo signal source
USB-GPIB adapter
High voltage probe

Rack Mount Kit
Bag

SPL2016
DF2001A
STB3
USB-GPIB
HPB4010
DPB1300/DPB4080/DPB5150/DPB5150A/DPB570
0/DPB5700A
CPL5100/CP4020/CP4050/CP4070/CP4070A/CP6
030/CP6030A/CP6150/CP6500
SDS2000HD-RMK
BAG-S2

Options

Part No.

Waveform generator (software)Power Analysis (software)
I2S trigger & decode (software)
MIL-STD-1553B trigger & decode (software)
FlexRay trigger & decode (software)
CAN FD trigger & decode (software)
SENT trigger & decode (software)
Manchester decode (software)
ARINC429 decode (software)
200 MHz to 350 MHz bandwidth upgrade (software)
200 MHz to 500 MHz bandwidth upgrade (software)
350 MHz to 500 MHz bandwidth upgrade (software)

SDS2000HD-FG
SDS2000HD-PA
SDS2000HD-I2S
SDS2000HD-1553B
SDS2000HD-FlexRay
SDS2000HD-CANFDSDS2000HD-SENT
SDS2000HD-Manch
SDS2000HD-ARINC
SDS2000HD-BW2T3
SDS2000HD-BW2T5
SDS2000HD-BW3T5

High voltage differential probe
Current probe

16

WWW.SIGLENT.COM

SDS2000X HD Series Digital Storage Oscilloscope

About SIGLENTSIGLENT is an international high-tech company, concentrating on R&D, sales,
production and services of electronic test & measurement instruments.

SIGLENT first began developing digital oscilloscopes independently in 2002.
After more than a decade of continuous development, SIGLENT has extended
its product line to include digital oscilloscopes, isolated handheld
oscilloscopes,

function/arbitrary

waveform

generators,

RF/MWsignalgenerators, spectrum analyzers, vector network analyzers, digital multimeters,
DC power supplies, electronic loads and other general purpose test
instrumentation. Since its first oscilloscope was launched in 2005, SIGLENT
has become the fastest growing manufacturer of digital oscilloscopes. We
firmly believe that today SIGLENT is the best value in electronic test &measurement.

Headquarters:
SIGLENT Technologies Co., Ltd
Add: Bldg No.4 & No.5, Antongda Industrial
Zone, 3rd Liuxian Road, Bao'an District,
Shenzhen, 518101, China
Tel: + 86 755 3688 7876
Fax: + 86 755 3359 1582
Email: sales@siglent.com
Website: int.siglent.com

North America:
SIGLENT Technologies America, Inc
6557 Cochran Rd Solon, Ohio 44139
Tel: 440-398-5800
Toll Free: 877-515-5551
Fax: 440-399-1211
Email: info@siglent.com

Follow us on
Facebook: SiglentTech

Website: www.siglentna.com

Europe:SIGLENT Technologies Germany GmbH
Add: Staetzlinger Str. 70
86165 Augsburg, Germany
Tel: +49(0)-821-666 0 111 0
Fax: +49(0)-821-666 0 111 22
Email: info-eu@siglent.com
Website: www.siglenteu.com

WWW.SIGLENT.COM

17