SSA3000X-R
Real-Time
Spectrum Analyzer
DataSheet

DS0703R_E02F

SSA3000X-R Analyzer DataSheetGeneral Description
The SIGLENT SSA3000X-R real-time spectrum analyzers are powerful and flexible tools for complex RF
spectrum, signal analysis and network analysis.
With a capability of 40 MHz analysis bandwidth and 7.2 µs 100% POI, the analyzer can provide
multi-dimensions data displays, advanced triggering, and RF data capturing, to solve modern RF
spectrum challenges, like hopping frequency, conflict channel, spectrum interference, and analog/digitalmodulation analysis, EMI pre-compliance test. They also provide a 1-path-2-port vector network analyzer
and a distance-to-fault locator for S-parameter measurement, cable and antenna testing.
Applications include broadcast monitoring/evaluation, cellular site, IoT, Wlan and Bluetooth, surveying,
research and development, education, production, and maintenance.

Features and Benefits
Spectrum Analyzer Frequency Range from 9 kHz up to 7.5 GHz
Vector Network Analyzer Frequency Range from 100 kHz up to 7.5 GHz-165 dBm/Hz Displayed Average Noise Level (Typ.)-98 dBc/Hz.@10 kHz Offset Phase Noise (1 GHz, Typ.)Level Measurement Uncertainty < 0.7 dB (Typ.)1 Hz Minimum Resolution Bandwidth (RBW)
Preamplifier and Tracking Generator Standard
Up to 40 MHz Real Time Analysis Bandwidth (Opt.)
100% POI 7.20 μs, Dynamic Range 60 dB, Multi-view for Density, Spectrogram, PvT and 3D
Distance To Fault
Advanced Measurement Kit (Opt.)
Modulation Analysis Mode (Opt.)
EMI Measurement Mode (Opt.)
10.1 inch Multi-Touch Screen , Mouse and Keyboard supported
Web Browser Remote Control on PC and Mobile Terminals and File Operation

Real-Time Spectrum Analyzer DataSheet

1Models and Main index
Model

SSA3032X-R

SSA3050X-R

SSA3075X-R

Frequency Range

9 kHz~3.2 GHz

9 kHz~5.0 GHz

9 kHz~7.5 GHz

Resolution Bandwidth

1 Hz~3 MHz

1 Hz~3 MHz

1 Hz~3 MHz

Displayed Average Noise Level-165 dBm/Hz-165 dBm/Hz-165 dBm/Hz

SSB Phase Noise<-98 dBc/Hz<-98 dBc/Hz<-98 dBc/Hz

Third-order intercept(TOI)+14 dbm+14 dbm+14 dbm

Total Amplitude Accuracy< 0.7 dB

< 0.7 dB

< 0.7 dB

Tracking Generator

100 kHz - 3.2 GHz

100 kHz - 5.0 GHz

100 kHz - 7.5 GHz

Real Time Band Width

25 MHz, 40 MHz (Option)

RTSA SFDR

60 dB

100% POI

7.20 μs

RTSA Measurement

Density, Spectrogram, 3D, PvT

VNA measurement

Vector S11, Vector S21

VNA Dynamic Range

90 dB

Distance to Fault

Timing Domain Analysis Locator

Touch Screen

Multi Touch, Mouse and Keyboard supported

Advanced Measurement

CHP, ACPR, OBW, CNR, Harmonic, TOI, Monitor

Modulation Analysis

AM, FM, PM, ASK, FSK, MSK, PSK, QAM

EMI Measurement

EMI Filter and Quasi-Peak Detector, Log Scale and Limit Line

Communication Interface

LAN, USB Device, USB Host (USB-GPIB)

Remote Control Capability

SCPI/Labview/IVI based on USB-TMC/VXI-11/Socket/Telnet

Remote Controller

NI-MAX, Web Browser, Easy Spectrum software, File Explorer

2Real-Time Spectrum Analyzer DataSheetDesign Features
Spectrum Analyzer Mode
10.1 lnch Display with Multi-Touch Screen

Phase noise <-98 dBc/Hz@1 GHz

Minimum 1 Hz Resolution Bandwidth (RBW)ACPR in Advanced Measurement Kit-165 dBm/Hz Displayed Average Noise Level

Monitor in Advanced Measurement Kit

Real-Time Spectrum Analyzer DataSheet

3Real Time Analysis Mode
Density,3D,Spectrogram,PvT,Multi-view and dimensions to monitor complex signals

Modulation Analysis Mode
AM/FM/PM, ASK/FSK/PSK/MSK/QAM Vector Signal Modulation Analysis and EVM evaluation, and Data
recording to PC. The analysis BW is same with real-time BW in RTSA mode

4

Real-Time Spectrum Analyzer DataSheetVector Network Analyzer Mode
100k-7.5GHz Vector S11 and S21 measurement, Multi Formats Overlay Display

Distance To Fault Mode
Cable and Antenna Test based on Timing Domain Analysis

Real-Time Spectrum Analyzer DataSheet

5EMI Measurement Mode
EMI Measurement with CISPR 16-1-1 EMI filter, Quasi-peak Detector, and pre-stored standards.

Accessories

6

Utility Kit

Near Field Probe Set

USB-GPIB Adaptor

6U Rack Mount

Soft Carrying Bag

Calibration Kit

Real-Time Spectrum Analyzer DataSheet

Specifications

Specifications are valid under the following conditions: The instrument is within the calibration period, hasbeen stored between 0 and 50°C for at least 2 hours prior to use, and has been powered on and warmed
up for at least 40 minutes. The specifications include the measurement uncertainty, unless otherwise
noted.

Specifications: All products are guaranteed to meet published specifications when operating at room
temperature (approximately 25°C), unless otherwise noted.

Typical: Performance deemed typical implies that 80 percent of the measurement results will meet the
typical published performance with a 95th percentile confidence level at room temperature (approximately
25°C). Typical performance is not warranted and does not include measurement uncertainty.

Nominal: The expected performance or design attribute.

Real-Time Spectrum Analyzer DataSheet

7

Spectrum Analyzer Mode
Frequency and Time Characteristic
Frequency
SSA3032R

SSA3050X-R

SSA3075X-R

Frequency range

9 kHz~3.2 GHz

9 kHz~5.0 GHz

9 kHz~7.5 GHz

Frequency resolution

1 Hz

Frequency Span
Range

0 Hz, 100 Hz to Max Frequency

Accuracy± Span / (number of display points - 1)

Internal Reference Source
Reference frequency

10.000000 MHz

Reference frequency accuracy /±[(time since last adjustment ×frequency aging rate) + temperature

uncertainty

stability + initial calibration accuracy]

Initial calibration accuracy<1 ppmTemperature stability<1 ppm, 0 °C ~50 °C

Frequency aging rate

<0.5 ppm/first year, 3.0 ppm/20 years

MarkerMarker resolution
Marker uncertainty
Frequency Counter resolution

Span / (number of display points - 1)
± [frequency indication × reference frequency uncertainty + 10% ×
resolution bandwidth + ½ * marker resolution + 1 Hz]
0.1 HzBandwidths
Resolution bandwidth (-3dB)

1 Hz ~ 3 MHz, in 1-3-10 sequence

Resolution filter shape factor< 4.8 : 1 (60 dB:3 dB), Gaussian-like

RBW uncertainty< 5%Video bandwidth (-3dB)

1 Hz ~ 10 MHz, in 1-3-10 sequence

VBW uncertainty< 5%

Sweep and Trigger
Sweep time
Sweep mode

1 ms to 7500 s
RBW = 3k Hz ~ 3 MHz, Sweep
RBW = 1 Hz ~ 10 kHz, FFT

Sweep rule

Single, Continuous

Trigger source

Free, Video, External

External trigger

5V TTL level, Rising edge/Falling edge

8

Real-Time Spectrum Analyzer DataSheetAmplitude Accuracy and Range Specifications
Amplitude and Level
Measurement range

DANL to +10 dBm, 100 kHz ~ 1 MHz, Preamp off
DANL to +20 dBm, 1 MHz ~ 7.5 GHz, Preamp off

Reference level-200 dBm to +30 dBm, 1 dB steps

Preamplifier

20 dB (nom.)

Input attenuation

0 ~ 50 dB, 1 dB steps

Maximum input DC voltage+/- 50 VDC

Maximum average power

30 dBm, 3 minutes, fc ≥10 MHz, att > 20 dBm, preamp off

Maximum damage level

33 dBm, fc≥10 MHz, att>20 dBm, preamp off

Level Display
Logarithmic level axis

1 dB to 200 dB

Linear level axis

0 to reference level, 0% to 100%

Units of level axis

dBm, dBmV, dBμV, dBμA, Volt, Watt

Number of display points

751

Number of traces

4

Trace detectors
Trace functions

Positive-peak, Negative-peak, Sample, Normal,
Average(Voltage/RMS/Video)
Clear write, Max Hold, Min Hold, View, Blank, Average, Math

SSB Phase Noise
Offset
10 kHz
100 kHz
1 MHz

20 °C to 30 °C, fc = 1 GHz, Normalized to 1 Hz
-96 dBc/Hz,-98 dBc/Hz (typ.)-95 dBc/Hz,-97 dBc/Hz (typ.)-112 dBc/Hz,-114 dBc/Hz (typ.)Real-Time Spectrum Analyzer DataSheet

9Displayed Average Noise Level (DANL)SSA3032X-R

SSA3050X-R

SSA3075X-R

20 °C to 30 °C, att = 0 dB, RBW = 1 Hz, sample detector, trace average > 50, TG off
100 kHz ~1 MHz
1 MHz~10 MHz
10 MHz~200 MHz
Preampoff

200 MHz~1.5 GHz
1.5 GHz~3.2 GHz-105 dBm,-105 dBm,-105 dBm,-109 dBm (typ.)-109 dBm (typ.)-109 dBm (typ.)-122 dBm,-122 dBm,-122 dBm,-126 dBm (typ.)-126 dBm (typ.)-126 dBm (typ.)-142 dBm,-142 dBm,-142 dBm,-146 dBm (typ.)-146 dBm (typ.)-146 dBm (typ.)-142 dBm,-142 dBm,-142 dBm,-147 dBm (typ.)-147 dBm (typ.)-147 dBm (typ.)-140 dBm,-140 dBm,-140 dBm,-145 dBm (typ.)-145 dBm (typ.)-145 dBm (typ.)-137 dBm,-137 dBm,-143 dBm (typ.)-143 dBm (typ.)3.2 GHz~5.0 GHz
5.0 GHz~6.5 GHz-136 dBm,-141 dBm (typ.)6.5 GHz~7.5 GHz-131 dBm,-139 dBm (typ.)100 kHz ~1 MHz
1 MHz~10 MHz
10 MHz~200 MHz
200 MHz~1.5 GHz
Preampon

1.5 GHz~3.2 GHz
3.2 GHz~5.0 GHz
5.0 GHz~6.5 GHz-133 dBm,-133 dBm,-133 dBm,-136 dBm (typ.)-136 dBm (typ.)-136 dBm (typ.)-151 dBm,-151 dBm,-151 dBm,-154 dBm (typ.)-154 dBm (typ.)-154 dBm (typ.)-161 dBm,-161 dBm,-161 dBm,-165 dBm (typ.)-165 dBm (typ.)-165 dBm (typ.)-159 dBm,-159 dBm,-159 dBm,-163 dBm (typ.)-163 dBm (typ.)-163 dBm (typ.)-159 dBm,-159 dBm,-159 dBm,-162 dBm (typ.)-162 dBm (typ.)-162 dBm(typ.)-157 dBm,-157 dBm,-161 dBm (typ.)-161 dBm (typ.)-157 dBm,-160 dBm (typ.)6.5 GHz~7.5 GHz-155 dBm,-159 dBm (typ.)10

Real-Time Spectrum Analyzer DataSheetFrequency Response
20 °C to 30 °C, 30% to 70% relative humidity, att = 20 dB, relative to 50 MHz
Preamp off±0.8 dB, ±0.4 dB (typ.)Preamp on±1.2 dB, ±0.6 dB (typ.)Error and Accuracy
Resolution bandwidth

Logarithmic resolution, relative to RBW = 10 kHz

switching uncertainty± 0.2 dB (nom.)

Input attenuation

20 °C to 30 °C, fc = 50 MHz, preamp off, relative to att = 20 dB

switching uncertainty± 0.5 dB
20 °C to 30 °C, fc = 50 MHz, RBW = VBW = 1 kHz, att = 20 dB, peak

Absolute amplitudedetector, 95% reliability

accuracy±0.4 dB, input signal -20 dBm, Preamp off±0.6 dB, input signal -40 dBm, Preamp on
20 °C to 30 °C, fc>100 kHz, input signal -50 dBm ~ 0 dBm, att = 20 dB,

Total amplitude accuracy

RBW=VBW=1 kHz, peak detector, preamp off, 95% reliability±0.7 dB

RF input VSWR

Att = 10 dB, 1 MHz~7.5 GHz<1.5 (nom.)

Distortion and Spurious Responses
Second harmonic

20 °C to 30 °C, fc ≥ 50 MHz, mixer level -20 dBm, att = 0 dB, preamp off

distortion (SHI)-65 dBc / +45 dBm (nom.)

Third-order intercept(TOI)
1dB gain compression
Residual response
Input related spurious

20 °C to 30 °C, fc ≥ 50 MHz, two -20 dBm tones spaced by 100 kHz, att = 0
dB, preamp off+14 dBm (typ.)20 °C to 30 °C, fc ≥ 50 MHz, att = 0 dB, preamp off> 0 dBm (nom.)
20 °C to 30 °C, input terminated = 50 Ω, att = 0 dB
< -90 dBm
20 °C to 30 °C, mixer level = -30 dBm<-65 dBc

Real-Time Spectrum Analyzer DataSheet11Tracking Generator
Frequency Parameter
Frequency Range

SSA3032X-R

SSA3050X-R

SSA3075X-R

100 kHz ~ 3.2 GHz

100 kHz ~ 5.0 GHz

100 kHz ~ 7.5 GHz

Frequency Resolution

1 Hz, Zero Span

RBW, sweep mode

3 kHz ~ 3 MHz

Power Parameter
Output level-40 dBm ~ 0 dBm

Output level resolution

1 dB

Output flatness+/-3 dB (nom.)

Normalization Trace

Ref A/B/C/D->Ref trace

VSWR< 2 (nom.)

Connector and ImpendenceN-type female, 50 Ω

Average safe reverse power

Total : 30 dBm (1 W)

Maximum safe reverse level

Voltage: ±50 VDCAdvanced Measurement Kit
Power Measurement
CHP, Channel Power

Channel Power, Power Spectral Density

ACPR, Adjacent Channel Power

Main CH Power, Left channel power, Right channel power

Ratio
OBW, Occupied Bandwidth

Occupied Bandwidth, Transmit Frequency Error

T-Power, Time Domain Power

Zero Span Integrated Power

CNR, Carrier Noise Ratio

C/N, Noise Power

Non-Linear Measurement
Harmonic measurement

Max Harmonic number 10

TOI, Third-Order Intercept

Measure the third-order products from two tones

Spectrum Monitor Measurement
Spectrogram

12

Real-Time Spectrum Analyzer DataSheetReal-Time Spectrum Analyzer Mode
Frequency and Time
Real-Time Bandwidth

25 MHz (Default)
40 MHz (Option SSA3000XR-RT40)

100% POI Minimum

Full Span, Kaiser Window, Frequency Mask Triggering at full amplitude accuracy

Signal Duration

7.20 μs

Measurement view

Density

30 ms ~ 50 s

3D+Spectrogram

30 ms ~ 50 s

Spectrogram

100 us ~ 50 s

PvT+Spectrum

100 us ~ 50 s

Points800

MAX Sample rate

51.2 MHz

FFT

150 000(40 MHz analysis BW)

Marker8

Span min

5 kHz

Window

Kaiser(Default), Hanning, Flattop, Gaussian, Blackman-Harris, Rectangular
Any SPAN, six RBW for every window (only one for Rectangular), default min RBW.
Typical RBW for Kaiser：

RBW

Spectrogram / PvTMaximum stored

Span

RBW min

RBW MAX

40 MHz100.43 kHz

3.3142 MHz

20 MHz50.21 kHz1.657 MHz

10 MHz25.11 kHz828.55 kHz

1 MHz

2.51 kHz82.85 kHz

100 kHz251 Hz8.285 kHz

50 000 (Loop store)

Different RBW and span, 100% POI (μs)
Analysis BW

RBW1

RBW2

RBW3

RBW4

RBW5

RBW6

40 MHz

26.56

16.56

11.56

9.06

7.81

7.20

20 MHz

46.56

26.56

16.56

11.56

9.06

7.81

10 MHz

86.56

46.56

26.56

16.56

11.56

9.06

1 MHz

806.56406.56206.56106.56

56.56

31.56

Real-Time Spectrum Analyzer DataSheet

13

Different window length for RBW
Length\Type

1024

512

256

128

64

32

Kaiser(Beta=12)398.2849

198.9478

99.279349.4450

24.5279

12.0693

Hanning

533.4785

266.4785

132.9785

66.2285

32.8535

16.1660

Flattop212.2447

106.0182

52.9050

26.3483

13.0700

6.4309Gaussian(alpha=3.5)

404.8707

202.2399

100.9244

50.2666

24.9376

12.2729

Blackman-Harris399.2401

199.4250

99.5174

49.5636

24.5868

12.0983

Rectangular801

400.5000

200.2500

100.1250

50.062525.0313

Amplitude Accuracy and Range
Detector+Peak, -Peak, Sample, Average

Trace

3

Spectrum Density Display

0~100% (resolution 0.1%)

Dynamic range forSpectrogram
Amplitude
TriggerFrequency Mask Trigger(FMT)
Colour Mode

14

200 dB
Flatness< 0.4 dB

Resolution

0.01 dB

Dynamic range< 60 dB

Free Run, PvT, External
SourceTraces

Type

Greater Than, Less Than, Outside Mask, Inside Mask

Actions

Stop, BeepWarm(Default), Cool, Gray

Real-Time Spectrum Analyzer DataSheetModulation Analyzer Mode
Common Parameter
SSA3032X-R

SSA3050X-R

SSA3075X-R

Frequency Range

2 MHz ~ 3.2 GHz

2 MHz ~ 5.0 GHz

2 MHz ~ 7.5 GHz

Carrier Power Accuracy±2 dB (nom.)

Carrier Power Range-30 dBm to +20 dBm (nom.)

Analog Modulation Analysis
AM
Modulation rate range
Accuracy

20 Hz to 100 kHz
1 Hz (nom.)

Modulation rate < 1 kHz< 0.1% modulation rate (nom.)

Modulation rate ≥ 1 kHz

Modulation depth range

5% to 95%

Accuracy±4% (nom.)FM
Modulation rate range
Accuracy

20 Hz to 200 kHz
1 Hz (nom.)

Modulation rate < 1 kHz< 0.1% modulation rate (nom.)

Modulation rate ≥ 1 kHz

Frequency deviation

1 kHz to 400 kHz

Accuracy±4% (nom.)PM
Modulation rate range
Accuracy

50 Hz ~ 50 kHz
1 Hz (nom.)

Modulation rate < 1 kHz< 0.1% modulation rate (nom.)

Modulation rate ≥ 1 kHz

Phase deviation

0.2 ~ 100 rad

Accuracy±4% (nom.)

Real-Time Spectrum Analyzer DataSheet

15

Digital Modulation Analysis
Measurement
ASK: 2ASK;
Modulation TypeFSK: 2,4,8,16 level;(The analysis BW is same

MSK: GMSK;

with real-time BW in RTSA

PSK: BPSK,QPSK,OQPSK,8PSK;

mode)

DPSK: DBPSK, DQPSK, D8PSK, π/4 -DQPSK, π/8 -D8PSK;
QAM: 16,32,64,128,256

Meas Length

16 to 4096

Points/Symbol

4,6,8,10,12,14,16

Symbol Rate

1 ksps to 25 Msps, Symbol Rate* Points/Symbol <=150 Msps

FilterMeas/Ref Filter

Nyquist, Squrt Nyquist, Gauss, Half Sine, Rectangular

Length2 to 128

Alpha/BT

Alpha 0.01~1, BT 0.01~10

Trace
IQ Meas Time, IQ Meas Spectrum,
IQ Ref Time, IQ Ref Spectrum,
Trace Data

Time, Spectrum,
Symbol Error Chart, Err Vector Time, Err Vector Spectrum,
IQ Mag Err, IQ Phase Err

LayoutSingle, Stacked 2, Grid 1 2, Grid 2*2
Log mag, Lin mag, Real, Imag,

Trace Formats

I-Q, Constellation, I-eye, Q-eye,Wrap Phase, Unwrap Phase, Trellis eye

Symbol Error Chart
EVM (rms EVM, peak EVM), Magnitude error,
PSK/DPSK/MSK/QAM

Phase error, IQ offset, Carrier offset, SNR Quadrature error,
Gain imbalance(not support for MSK)ASKASK Error, ASK depth, carrier offset

FSK

FSK Error, Magnitude error, FSK deviation, carrier offset

16

Real-Time Spectrum Analyzer DataSheetVector Network Analyzer Mode
Stimulus and Measurement
SSA3032X-R

SSA3050X-R

SSA3075X-R

Frequency Range

100 kHz ~ 3.2 GHz

100 kHz ~ 5.0 GHz

100 kHz ~ 7.5 GHz

Measurement

S11, S21

IFBW

10 kHzPort1 Stimulus Power-20 dBm ~ 0 dBm-20 dBm ~ 0 dBm-40 dBm ~ 0 dBm

Lin Mag, Log Mag, Phase, Group Delay, SWR,
FormatSmith Chart (Lin/Phase, Log/Phase, Real/Imag, R+j*X, G+j*B),
Polar Chart (Lin/Phase, Log/Phase, Real/Imag)

Sweep Points

101~10001, default 201

Trace

4 traces, Mem, Math, Hold, OverlayMarker(6+Ref)* 4 traces

Calibration
Directivity of Calibration

S11, Log mag, Average=50, >50MHz> 40 dB
S21, IFBW=10 kHz, Port1 level=-5 dBm, Log Mag, Average=50

Dynamic
Range

100 kHz ~ 10 MHz

60 dB (typ.)60 dB (typ.)60 dB (typ.)10 MHz ~ 1.5 GHz

90 dB (typ.)90 dB (typ.)90 dB (typ.)1.5 GHz ~ 3.2 GHz

90 dB (typ.)90 dB (typ.)90 dB (typ.)80 dB (typ.)80 dB (typ.)3.2 GHz ~ 7.5 GHz
Trace Noise

10 kHz RBW, Log mag, Average = 50, >10MHz
0.1 dB
Short Response
Open Response

Calibration Type

Full 1-Port(OSL)
Response ThroughEnhanced Response

Mechanical Calibration Kit

F503ME, F503FE, F603ME, F603FE, F504MS, F504FS, F604MS, F604FS,
85032B\E, 85033E, 85032F, User Cal Kit

Port Extensions

Port 1, Port 2, Auto Open Port 1

System Z0

50 Ω

Velocity Factor

0.1~1

Real-Time Spectrum Analyzer DataSheet

17

Distance to Fault Mode
Measurement
SSA3032X-R

SSA3050X-R

SSA3075X-R

Frequency Range

100 kHz ~ 3.2 GHz

100 kHz ~ 5.0 GHz

100 kHz ~ 7.5 GHz

Maximum Distance (meters)(76800 x Velocity Factor)/(Stop Freq - Start Freq (MHz))

Resolution (meters)(150 x Velocity Factor)/(Stop Freq - Start Freq (MHz))Windows

Rectangular, Hamming

Calibration

Full 1-Port (OSL)

Velocity Factor

0.1~1

EMI Measurement Mode
Measurement
Measurement View

Frequency scan, Meter, Signal list

Pre-compliance Sequence

Scan, Search, Meas

EMI filter RBW (-6dB)

200 Hz, 9 kHz, 120 kHz, 1MHz(following CISPR 16-1-1)

RBW uncertainty< 5% (nom.)

Detector

Peak, Voltage Average, Quasi-Peak(following CISPR 16-1-1)

Dwell time

0 us ~ 10 s

RBW/Steps

0.1, 0.3, 0.5, 1, 2, 3

Corrections

4

Limit and Trace

3

Limit Standards

EN550xx, GB9254, FCC Part15, User defined

Attenuator

0-50 dB

ReportSignal List

Frequency scale

Linear, Logarithmic

18

Real-Time Spectrum Analyzer DataSheetInputs and Outputs
Front Panel
RF input, Port 2

N-type female, 50 Ω (nom.)

TG Source, Port 1

N-type female, 50 Ω (nom.)

USB host

USB-A plug, version 2.0

Ear Phone Jack

3.5 mmRear PanelUSB device

USB-B plug, version 2.0

LAN10/100 Base, RJ-45

10 MHz reference output

10 MHz, >0 dBm, BNC-type female, 50 Ω (nom.)

10 MHz reference input

10 MHz, -5 to +10 dBm, BNC-type female, 50 Ω (nom.)

External trigger input

5V TTL level, BNC-type female, 10 kΩ

Remote Control
Communication Interface

LAN, USB Device, USB Host (USB-GPIB adaptor)
SCPI / Labview / IVI based on USB-TMC / VXI-11 / Socket / Telnet;
NI-MAX;

Remote Control Capability

Web Browser (HTML 5 Supported);
Easy Spectrum software;
File Explorer (FTP)

Real-Time Spectrum Analyzer DataSheet

19General Specification
Structure
Dimensions

393 mm × 207 mm × 116.5 mm (W×H×D)

Weight

Net: 4.70 kg (10 lb); Shipping: 5.50 kg

Display

TFT LCD, 1024 × 600, 10.1 inch capacitive multi-touch screen

Storage

Internal (Flash) 256 MB, external (USB storage device) 32 GB

Working Environment
SourceTemperature
Humidity
AltitudeAC voltage range: 100-240 V, 50/60 Hz or 100-120 V 400 Hz;
Power consumption: 70 W (MAX)Working temperature: 0 °C to 40 °C,
Storage temperature: -20 °C to 70 °C
0°
C to 30 °C, ≤ 95% Relative humidity
30 °C to 50 °C, ≤ 75% Relative humidity
Operating: less than 3 km

Electromagnetic Compatibility
EN 61326-1: 2013 /Class A(The active input power of the EUT is less than 75 W. According to

EN 61000-3-2: 2014

EN 61000-3-2, no limits are necessary.)

EN 61000-3-3: 2013

Plt: 0.65 Pst: 1.00, dmax: 4.00 %, dc: 3.00 %;dt Lim: 3.30 % dt>Lim: 500ms

IEC 61000-4-2: 2008

AD ±8.0 kV, CD ±4.0 kV

IEC 61000-4-3: 2006 +80 MHz to 1000 MHz: 10V/m, 1.4 GHz to 2.0 GHz:3 V/m,

A1: 2007 + A2: 2010

2.0 GHz to 2.7 GHz:1V/m

IEC 61000-4-4: 2004 +A1: 2010

AC Line:±2.00 kV

IEC 61000-4-5: 2005

Line to Line: 1.0 kV, Line to Earth: 2.0 kV

IEC 61000-4-6: 2008

0.15-80 MHz:3 V 1 KHz 80% AM

IEC 61000-4-8: 200930 A/m, 50/60 Hz

IEC 61000-4-11: 2004

Voltage Dips:0%/0.5P; 40%/10P; 70%/25P;
Short Interruptions Test Level % UT: 0%/250P

SafetyIEC 61010-1:2010/EN 61010-1:2010
CAN/CSA-C22.2 No.61010-1:2012,
CAN/CSA-C22.2 No.61010-2-30:2012,
UL 61010-1:2012,
UL 61010-2-30:2012
RoHS
2011/65/EU

20

Real-Time Spectrum Analyzer DataSheetOrdering Information
Product

Product CodeDescription

Order Number

Real Time Spectrum Analyzer, 9 kHz ~ 3.2 GHz, Preamp
and TG standard, VNA standard

SSA3032X-R

Real Time Spectrum Analyzer, 9 kHz ~ 5.0 GHz, Preamp
and TG standard, VNA standard
Real Time Spectrum Analyzer, 9 kHz ~ 7.5 GHz, Preamp
and TG standard, VNA standard

Standard Accessories

SSA3050X-R
SSA3075X-R

Quick Start, USB Cable, Power Cord
Advanced Measurement Kit

SSA3000XR-AMK

40 MHz Analysis BandWidth

SSA3000XR-RT40

Utility Kit:
N(M)-SMA(M) cable(6 GHz), N(M)-N(M) cable(6 GHz),
N(M)-BNC(F) adaptor x2, N(M)-SMA(F) adaptor x2,

UKitSSA3X

10 dB 1W attenuator
Common Options andAccessories

VNA Options

EMI Measurement
Options

N(M)-BNC(M) cable, 70cm, 2 GHz

N-BNC-2L

N(M)-SMA(M) cable, 70cm, 6 GHz

N-SMA-6L

N(M)-N(M) cable, 70cm, 6 GHz

N-N-6L

N(M)-SMA(M) cable, 100cm, 18 GHz

N-SMA-18L

N(M)-N(M) cable, 100cm, 18 GHz

N-N-18L

SMA(M)-SMA(M) cable, 100cm, 18 GHz

SMA-SMA-18L

USB-GPIB Adaptor

USB-GPIBSoft carrying bag

BAG-S2

6U Rack Mount Kit

SSA-RMKN type Economic Calibration Kit, DC~4.5GHz, 50 Ω

F503MEN type Economic Calibration Kit, DC~4.5GHz, 50 Ω

F503FE

3.5mm type Economic Calibration Kit, DC~4.5GHz, 50 Ω

F603ME

3.5mm type Economic Calibration Kit, DC~4.5GHz, 50 Ω

F603FEN type Standard Calibration Kit, DC~9GHz, 50 Ω

F504MSN type Standard Calibration Kit, DC~9GHz, 50 Ω

F504FS

3.5mm type Standard Calibration Kit, DC~9GHz, 50 Ω

F604MS

3.5mm type Standard Calibration Kit, DC~9GHz, 50 Ω

F604FSEMI Measurement Mode

SSA3000XR-EMI

300 kHz~3 GHz Near Field Probe Kit:
3 H-probes (20/10/5 mm), 1 E-probe (5 mm)

SRF5030T

Modulation Analysis

Analog Modulation Analysis: AM, FM, PM

SSA3000XR-AMA

Options

Digital Modulation Analysis: ASK, FSK, MSK, PSK, QAM

SSA3000XR-WDMA

Real-Time Spectrum Analyzer DataSheet21