SHF Communication Technologies AGWilhelm-von-Siemens-Str. 23D • 12277 Berlin • Germany
Phone ++49 30 / 772 05 10 • Fax ++49 30 / 753 10 78
E-Mail: sales@shf.de • Web: http://www.shf.dePreliminary
Datasheet
SHF 11110 A
SHF 11122 AError Analyzers

SHF reserves the right to change specifications and design without notice – 11110A / 11122A - V002 – 08.06.2009

Page 1/3

Description
The SHF 11110 A and SHF 11122 A are error analyzers for applications where cost, space andfunctionalities need to be carefully balanced. They are particularly suited for 40G components, module andsubsystem production tests.
1The units contain a built-in frequency synthesizer and clock recovery to access bit rates from 39.8 to 43.1
Gbps; therefore rendering them equally valuable as a general purpose error analyzer for a wide range ofdigital test applications covering the key bit rates from 39.8 to 43.1 Gbps.
79111520

23

The devices allow the analysis of PRBS signals with pattern lengths of 2 -1, 2 -1, 2 -1, 2 -1, 2 -1, 2 -1
31
and 2 -1.
The units are controlled over a standard Ethernet connection by an external computer. An easy to use
software package provides not only a user friendly interface for changing the operating parameters but
also the capabilities of feature enhancement through firmware & software upgrades.
The plug-in version SHF 11110 A is to be used together with the SHF 10000 B and SHF 10001 A
mainframes to allow an individual test setup together with modules from the broad selection of SHF 10000
series extension modules. For production testing up to 4 SHF 11110 A can be combined into one
mainframe for multiple device testing.
The stand alone bench top unit SHF 11122 A is a small size unit to be used in the case that no further
extension modules are required.

Feature1•

Differential data input interface (single ended operation supported)•

Built-in clock recovery•

Seven built-in PRBS patterns: 2 -1, 2 -1, 2 -1, 2 -1, 2 -1, 2 -1, 2 -1•

Supports bit rates of 39.8 to 43.1•

Sub-rate clock outputs (1/2 clock, 1/4 clock, 1/32 clock)•

Built-in frequency synthesizer for all bit rate operations (to provide the reference clock)•Supports external reference clock input•Internal gating by time and bits•

Remote operation by intuitive software interface or commonly used measurement control software

7911152023thThe unit can also be operated by using an external reference clock of 1/64 , 1/32
rate range.

31ndthor 1/16 of the 39.8 to 43.1 Gbps bit

SHF reserves the right to change specifications and design without notice – 11110A / 11122A - V002 – 08.06.2009

Page 2/3

Preliminary Specifications – SHF 11110A and SHF 11122 AParameter

UnitMin.

Typ.

Max.

CommentData Inputs
50 ΩConnector Type1.85 mm femaleBit rate

Gbps

39.8

43.1

Input levelmV

100800Single ended

Clock Input
Connector type
Reference Clock50 Ω

SMA female

Reference Clock frequency
Clock/32
Clock/64

MHzMHz1243.8621.91346.9673.4

Reference Clock Input level

mV

300

800

Vpp, internal AC coupled

Clock Output
50 ΩConnector type
Reference Clock output
Clock/2
Clock/4
Clock/32
Output level
Clock/2
Clock/4
Clock/32
Output frequency
Clock/2
Clock/4
Clock/32

SMA female
2.92 mm female
2.92 mm female
SMA femalemVGHzGHzMHz200
300
400

500
700
700

19.9
9.95
1243.821.5
10.71346.9Vpp
Ground-Referenced CML
Ground-Referenced CML
Ground-Referenced CML

Patterns
Standard CCITT PRBS

72 -1
92 -1
112 -1152 -1
20
2 -1
23
2 -1
31
2 -1

Apply to all bit rates

SHF reserves the right to change specifications and design without notice – 11110A / 11122A - V002 – 08.06.2009

Page 3/3