SDS6000A Series
Digital Storage
Oscilloscope

Data Sheet
2022.02

EN01D

Key Features
4 analog channels, up to 2 GHz bandwidth with 5 GSa/s(10 GSa/s ESR) sample rate at each channel

SDS6204A
SDS6104A
SDS6054A

Product Overview
SIGLENT’s SDS6000A series Digital Storage Oscilloscopes
are available in bandwidths of 2 GHz, 1 GHz and 500 MHz,
have sample rate of 5 GSa/s (10 GSa/s ESR) at each channel,
maximum record length of 500 Mpts/ch, and display up to 4
analog channels + 16 digital channels mixed signal analysis
ability.
The SDS6000A series employs Siglent’s SPO technology witha maximum waveform capture rate of up to 170,000 wfm/ s(normal mode, up to 750,000 wfm/s in Sequence mode), 256level intensity grading display function plus a color temperature
display mode. It also employs an innovative digital trigger
system with high sensitivity and low jitter. The trigger system
supports multiple powerful triggering modes including serial
bus triggering. Tools such as History waveform recording,Search and Navigate functions, Mask Test, Bode Plot, Power
Analysis and Eye/Jitter Analysis allow for extended waveformrecords to be captured, stored, and analyzed. An impressivearray of measurement and math capabilities, options for a 25
MHz arbitrary waveform generator, as well as serial decoding
are also features of the SDS6000A.
The large 12.1’’ display capacitive touch screen supports multitouch gestures, with the addition of user-friendly UI design, cangreatly improve the operation efficiency. It also supports mouse
control, and remote web control over LAN.

Low background noise, supports 0.5 mV/div to 10 V/div
vertical scales
SPO technology

Waveform capture rates up to 170,000 wfm/s(normal mode), and 750,000 wfm/s (sequence
mode)



Supports 256-level intensity grading and color
temperature display modes

500 Mpts Record length in total for all 4 channels

Digital trigger system

Intelligent trigger: Edge, Slope, Pulse, Window, Runt,
Interval, Dropout, Pattern, Qualified, Nth edge,
Setup/hold, Delay and Video (HDTV supported). Zone
Trigger simplifies advanced triggering
Serial bus triggering and decoder, supports protocols I2C,
SPI, UART, CAN, LIN, CAN FD, FlexRay, I2S, MIL-STD1553B, SENT, Manchester and ARINC429
Segmented acquisition (Sequence) mode, dividing the
maximum record length into multiple segments (up to
80,000), according to trigger conditions set by the user,
with a very small dead time between segments to capture
the qualifying event
History waveform record (History) function, the maximum
recorded waveform length is 80,000 frames
Automatic measurements on 50+ parameters, supports
statistics
withhistogram, track, trend, Gating
measurement, and measurements on Math, History andRef
4 Math traces (8 Mpts FFT, Filter, addition, subtraction,
multiplication, division, integration, differential, squareroot, etc.), supports formula editor
Abundant data analysis functions such as Search,
Navigate, Digital Voltmeter, Counter, Waveform
Histogram, Bode plot, Power Analysis and Eye/Jitter
AnalysisHigh Speed hardware-based Average, Hi-Res; High
Speed hardware-based Mask Test function, with Mask
Editor tool for creating user-defined masks
16 digital channels
25 MHz function / arbitrary waveform generator, built-in
multiple predefined waveforms
Large 12.1” TFT-LCD display with 1280 * 800 resolution;
Capacitive touch screen supports multi-touch gestures
Interfaces include: USB Hosts, USB Device (USBTMC),
LAN（VXI-11/Telnet/Socket）, micro SD card, Pass/Fail,
Trigger Out, HDMI
Built-in web server supports remote control over the LAN
port using a web browser. Supports SCPI remote control
commands. Supports external mouse and keyboard

WWW.SIGLENT.COM

1

SDS6000A Series Digital Storage Oscilloscope

Models and Key Specifications
Model

SDS6204A

SDS6104A

SDS6054A

Analog channels
Bandwidth
Sample rate
(Max.)

4 + EXT
2 GHz

1 GHz

500 MHz

Memory depth(Max.)

500 Mpts/ch（single-channel）
250 Mpts/ch (dual-channel)
125 Mpts/ch (3 or 4 channels)

Waveform capture
rate (Max.)

Normal mode：170,000 wfm/s;

Vertical resolution

8-bit, up to 16-bit in Hi-Res mode
Edge, Slope, Pulse width, Window, Runt, Interval, Dropout, Pattern, Video, Qualified, Nth edge,
Setup/hold, Delay, Serial

Trigger type

5 GSa/s (10 GSa/s ESR) @ each channel

Sequence mode：750,000 wfm/s

Serial trigger anddecodeStandard: I2C, SPI, UART, CAN, LIN
Optional: CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT, Manchester (decode only), ARINC429 (decode
only)Measurement

50+ parameters，statistics, histogram, trend, and track supported
4 traces

MathData analysis
Digital channel
Waveform
generator
I/O
Probe (Standard)
Display2

8 Mpts FFT, +, -, x, ÷, ∫dt, d/dt, √, Identity, Negation, Absolute, Sign, ex, 10x, ln, lg, Interpolation, MaxHold,
MinHold, ERES, Average, Filter. Supports formula editor
Search, Navigate, History, Mask Test, Digital Voltmeter, Counter, Waveform Histogram, Bode plot andPower Analysis, Eye/Jitter Analysis
16-channel; maximum sample rate up to 1 GSa/s; record length up to 50 Mpts
Single-channel external USB isolated waveform generator, frequency up to 25 MHz, 125 MSa/s sample
rate, 16 kpts waveform memory
USB 3.0 Host x2，USB 2.0 Host x2，USB 2.0 Device，LAN，micro SD card, HDMI, External trigger，
Auxiliary output（TRIG OUT，PASS/FAIL）
SP3150A, 500 MHz, 1 probe supplied for each channel
12.1 TFT-LCD with capacitive touch screen (1280*800)

WWW.SIGLENT.COM

SDS6000A Series Digital Storage Oscilloscope

Functions & Characteristics
Excellent User Interface and User Experience12.1” display with 1280*800 resolution

Capacitive touch screen, supporting multi-touchgestures, can move or scale the waveform traces
quickly by finger-touch movements, which greatly
improves the operation efficiency



Built-in WebServer supports remote control on a
web page over LAN

Supports external mouse and keyboard

High Waveform Update Rate

Deep Record LengthWith a waveform update rate of up to 170,000 wfm/s, theoscilloscope can easily capture unusual or low-probability
events. In Sequence mode, the waveform capture rate canreach 750,000 wfm/s

Using hardware-based Zoom technique and record length of up
to 500 Mpts, users can select a slower timebase withoutcompromising the sample rate, and then quickly zoom in tofocus on the area of interest

Multiple Trigger Functions

Trigger Zone

Edge, Slope, Pulse, Video, Windows, Runt, Interval, Dropout,
Pattern, Qualified, Nth edge, Setup/hold, Delay and serial
trigger

Trigger Zone is available for advanced triggering

WWW.SIGLENT.COM

3

SDS6000A Series Digital Storage Oscilloscope

Advanced Math Function

In addition to the traditional (+, -, X, /) operations, FFT, Filter,
integration, differential, square root, and more are supported.
Formula Editor is available for more complex operations. 4 mathtraces are available.

Hardware-accelerated FFT supports up to 8 Mpts operation.
This provides high-frequency resolution with a fast refresh rate.
The FFT function also supports a variety of window functions sothat it can adapt to different spectrum measurement needs.
Three modes (Normal, Average, and Max hold) can satisfy
different requirements for observing the power spectrum. Autopeak detection and markers are supported.

Measurements of a Variety of Parameters

Parameter Statistics Function

Parameter measurements include 4 categories: horizontal,
vertical, miscellaneous, and CH delay providing a total of 50+
different types of measurements. Measurements can be
performed within a specified gate period. Measurements on
Math, Reference, and History frames are supported

4

Statistics show the current value, maximum value, minimum
value, standard deviation, and mean value of up to 12
parameters simultaneously. A histogram is available to show the
probability distribution of a parameter. Trend and Track areavailable to show the parameter value vs. time.
For horizontal parameters such as period, all results are
extracted from a frame, instead of just calculating the first one.
This accelerates statistics on horizontal measurements much
more and enables distribution observation in a frame using
Histogram and Track

WWW.SIGLENT.COM

SDS6000A Series Digital Storage Oscilloscope

History Mode

Sequence Mode

History function can record up to 80,000 frames of waveforms.
The recording is executed automatically so that the customer
can playback the history waveforms at any time to observe
unusual events and quickly locate the area of interest using the
cursors or measurements. The failed frames of the Mask Test
can be stored as history

Segmented memory collection will store the waveform into
multiple memory segments (up to 80,000) and each segment
will store a triggered waveform as well the dead time
information. The interval between segments can be as small as
1.3 μs. All of the segments can be played back using the History
function

Search and Navigate

Serial Bus Decode

The oscilloscope can search events specified by the user in aframe. Events flagged by the Search can be recalled
automatically using Navigate. It can also navigate by time (delay
position) and history frames

Display the decoded characters through the events list. Bus
protocol information can be quickly and intuitively displayed intabular form. I2C, SPI, UART, CAN, LIN, CAN FD, FlexRay, I2S,
MIL-STD-1553B, SENT, Manchester and ARINC429 aresupported

WWW.SIGLENT.COM

5

SDS6000A Series Digital Storage Oscilloscope

Hardware-based High Speed Mask Test Function

The oscilloscope utilizes a hardware-based Mask Test function,
performing up to 18,000 Pass / Fail decisions each second. It iseasy to generate user-defined test templates to provide trace
mask comparisons, making it suitable for long-term signal
monitoring or automated production line testing

Built-in Mask Editor application helps to create custom masks

Eye/Jitter Analysis

Supports eye diagram and jitter analysis/measurement. It can automatically extract the embedded reference clock from serial dataand create the eye diagram. Measurement on multiple eye/jitter parameters is provided. Mask test on eye diagrams is supported

6

WWW.SIGLENT.COM

SDS6000A Series Digital Storage Oscilloscope

Bode PlotPower Analysis (Optional)

The oscilloscope can control the isolated USB AWG module ora stand-alone SIGLENT SDG generator, to scan the amplitude
and phase-frequency response of the DUT, and display the data
as a Bode Plot. This makes it possible to replace expensive
network analyzers in some applications

The Power Analysis option provides a full suite of power
measurements and analysis, which greatly improve themeasurement efficiency in switching power supplies and powerdevices design

Digital Channels / MSO

25 MHz Function/Arbitrary Waveform Generator

Four analog channels plus 16 digital channels enable users toacquire and trigger the waveforms then analyze the pattern,simultaneously with one instrument

The oscilloscope can control the SAG1021I isolated USB
Function/Arbitrary waveform generator to output waveform withup to 25 MHz frequency and ±3 V amplitude. Six basic
waveforms plus multiple types of arbitrary waveforms are builtin

Complete Connectivity

USB Host 3.0 x2, USB Host 2.0 x2, USB Device 2.0 (USBTMC) x1, LAN（VXI-11/Telnet/Socket）x1, micro SD card x1, Auxiliary
output（Pass/Fail, Trigger Out）x1 and HDMI x1

WWW.SIGLENT.COM

7

SDS6000A Series Digital Storage Oscilloscope

Specifications
All specifications are not guaranteed unless the following conditions are met:
• The oscilloscope calibration period is current• The oscilloscope has been working continuously for at least 30 minutes at the specified temperature (18℃ ~ 28℃ )Acquire (analog)
Sample rate

5 GSa/s (10 GSa/s (ESR*1) @ each channel

Memory depth *2

500 Mpts/ch（single-channel）
250 Mpts/ch (dual-channel)
125 Mpts/ch (3 or 4 channels)

Waveform update
rate

Normal mode：up to 170,000 wfm/s

Intensity grading
Peak detect
Average
Hi-ResSequence
History
Interpolation

256-level
200 ps4, 16, 32, 64, 128, 256, 512, 1024, 2048, 4096, 8192
Enhanced bit: 1, 2, 3, 4, 5, 6, 7, 8 bit
Up to 80,000 segments, interval between triggers = 1.3 μs min.
Up to 80,000 frames
sinx/x, x

Sequence mode：up to 750,000 wfm/s* 1：ESR： Enhanced Sample Rate，gets better measure accuracy by 2x interpolation* 2：In Average and Hi-Res modes，the memory depth is 25 Mpts/ch

Vertical(analog)

SDS6204A

SDS6104A

SDS6054A

2 GHz *1

1 GHz

500 MHz230 ps350 ps550 ps500 MHz

500 MHz

500 MHz

300 MHz

300 MHz

300 MHzChannelBandwidth(-3dB) @ 50Ω
Rise time@50Ω
(typical)Bandwidth(-3dB) @ 1 MΩ,with probe
Bandwidth(-3dB) @ 1 MΩ,with external 50Ω

4 + EXT

Resolution

Range
Vertical scale(probe 1X)
DC gain accuracy
Offset accuracy

8-bit, up to 16-bit in Hi-Res mode
9-bit: 0.25*Sample rate, up to the analog bandwidth
10-bit: 0.115*Sample rate, up to 1.15 GHz, limited by the analog bandwidth
11-bit: 0.055*Sample rate, up to 550 MHz, limited by the analog bandwidth
12-bit: 0.028*Sample rate, up to 280 MHz
13-bit: 0.014*Sample rate, up to 140 MHz
14-bit: 0.007*Sample rate, up to 70 MHz
15-bit: 0.0035*Sample rate, up to 35 MHz
16-bit: 0.0017*Sample rate, up to 17 MHz
8 divisions
1 MΩ: 0.5 mV/div – 10 V/div
50 Ω: 0.5 mV/div – 1 V/div±1.5%± (1% of the offset setting + 0.5% of full scale + 0.02% of max offset + 1mV)

Offset range(probe 1X)

1 MΩ: 0.5 mV/div ~ 5 mV/div: ± 1.6 V; 5.1 mV/div ~ 10 mV/div: ± 4 V; 10.2 mV/div ~ 20 mV/div : ± 8 V; 20.5
mV/div ~ 100 mV/div: ± 16 V；102 mV/div ~ 200 mV/div: ± 80V; 205 mV/div ~ 1 V/div: ± 160 V;1.02 V/div ~10 V/div: ± 400 V

Bandwidth in HiRes mode(typical)Bandwidth limit
Low frequency
response (ACcoupling -3 dB)
Overshoot (100mV/div, 150 ps
8

50 Ω: 0.5 mV/div ~ 5 mV/div: ± 1.6 V; 5.1 mV/div ~ 10 mV/div: ± 4 V; 10.2 mV/div ~ 20 mV/div: ± 8 V; 20.5
mV/div ~ 1 V/div: ± 10 V
Hardware Bandwidth limit: 20MHz, 200MHz
6 Hz (typical)
15%

10%

5%
WWW.SIGLENT.COM

SDS6000A Series Digital Storage Oscilloscopeedge @50 Ω,
typical)Coupling
Impedance
Max. Input
voltage
SFDR
CH to CH
Isolation (@50Ω)Probe Attenuation

DC, AC, GND(1 MΩ ± 2%) || (20 pF ± 3 pF)
50 Ω: 50 Ω ± 2%
1 MΩ ≤ 400 Vpk(DC + AC), DC ~ 10 kHz
50 Ω ≤ 5 Vrms, ± 10V Peak
≥ 45dBc
70 dB up to 200 MHz
60 dB up to 500 MHz
50 dB up to 1 GHz
40 dB up to 2 GHz
1X, 10X, 100X, custom* 1：The bandwidth is 1 GHz below 2.3 mV/div

Horizontal

SDS6204A

SDS6104A

SDS6054ATime scale
Range
Display mode
Roll mode
Skew (C1~C4)
Time base
Accuracy

0.1 ns/div – 1000 s/div
10 divisions
Y-T, X-Y, Roll
≥ 50 ms/div< 100 ps

0.2 ns/div – 1000 s/div

0.5 ns/div – 1000 s/div±2 ppm initial (0~50℃); ±0.5 ppm 1st year aging; ±3 ppm 20-year aging

TriggerMode

Auto, Normal, Single
Internal: ±4.5 div from the center of the screen

Level

Ext Trigger Channel
input voltage
Hold off range

Coupling

EXT: ± 0.61 V
EXT/5: ± 3.05 V
1 MΩ ≤ 42 Vpk
50 Ω ≤ 5 Vrms
By time: 8 ns ~ 30 s (8 ns step)
By event: 1 ~ 108
C1~C4
DC: Passes all components of the signal
AC: Blocks DC components and attenuates signals below 15 Hz
LFRJ: Attenuates the frequency components below 2.4 MHz
HFRJ: Attenuates the frequency components above 1.3 MHz
Noise RJ: Increases the trigger hysteresis
EXT
DC: Passes all components of the signal
AC: Blocks DC components and attenuates signals below 15 Hz
LFRJ: Attenuates the frequency components below 2.5 MHz
HFRJ: Attenuates the frequency components above 1.3 MHz

Accuracy (typical)

C1～C4: ±0.2 div
EXT: ±0.3 div
C1 ～ C4:

Sensitivity

EXT:

EXT/5:

JitterWWW.SIGLENT.COM>10 mV/div:
5 mV/div~10 mV/div:≤ 2 mV/div:

Noise RJ = OFF±0.26 div
±0.26 div±0.5 div

Noise RJ = ON
±0.33 div
±0.33 div±0.5 div

200 mVpp, DC ～ 10 MHz
300 mVpp, 10 MHz ～ bandwidth（300 MHz）
1 Vpp, DC ～ 10 MHz
1.5 Vpp, 10 MHz ～ bandwidth（300 MHz）

C1～C4:< 9 ps RMS (typical) for ≥ 300 MHz sine and ≥ 6 divisions peak to peak amplitude for vertical gain settingsfrom 2.5 mV/div to 10 V/div< 5 ps RMS (typical) for ≥ 500 MHz sine and ≥ 6 divisions peak to peak amplitude for vertical gain settingsfrom 2.5 mV/div to 10 V/div
EXT: < 200 ps rms
9

SDS6000A Series Digital Storage Oscilloscope
Displacement

Pre-Trigger: 0 ~ 100% memory
Delay-Trigger: 0 ~ 10,000 div
Up to 2 zones

Zone
Edge Trigger
SourceSlope
Slope Trigger
SourceSlope
Limit range

Source: C1~C4
Property: Intersect, Not Intersect
C1~C4/EXT/(EXT/5)/AC Line/D0~D15
Rising, Falling, Rising & Falling
C1~C4
Rising, Falling<, >, in range, out of range

Time range

2 ns ～ 20 s, Resolution = 1 ns

Pulse Width Trigger
SourcePolarity
Limit range

C1~C4/D0~D15+wid, -wid<, >, in range, out of range

Time range

2 ns ～ 20 s, Resolution = 1 ns

Video Trigger
SourceStandard
Synchronization
Trigger Condition
Window Trigger
SourceWindow typeInterval Trigger
SourceSlope
Limit range

C1~C4
NTSC, PAL, 720p/50, 720p/60, 1080p/50, 1080p/60, 1080i/50, 1080i/60, Custom
Any, SelectLine, Field
C1~C4
Absolute, Relative
C1~C4/D0~D15
Rising, Falling<, >, in range, out of range

Time range

2 ns ～ 20 s, Resolution = 1 ns

Dropout Trigger
SourceTimeout type
Slope

C1~C4/D0~D15
Edge, State
Rising, Falling

Time range

2 ns ～ 20 s, Resolution = 1 ns

Runt Trigger
SourcePolarity
Limit range

C1~C4
Positive, Negative<, >, in range, out of range

Time range

2 ns ～ 20 s, Resolution = 1 ns

Pattern Trigger
SourcePattern Setting
Logic
Limit range

C1~C4/D0~D15
Don’t Care, Low, HighAND, OR, NAND, NOR<, >, in range, out of range

Time range

2 ns ～ 20 s, Resolution = 1 ns

Qualified Trigger
Type
Qualified Source
Edge Trigger Source
Nth Edge Trigger
SourceSlope
Idle time

State, State with Delay, Edge, Edge with Delay
C1~C4/D0~D15
C1~C4/D0~D15
C1~C4/D0~D15
Rising, Falling
8 ns ～ 20 s, Resolution = 1 ns

Edge Number

1 ～ 65535

Delay Trigger
Source A
Source B

C1~C4/D0~D15
C1~C4/D0~D15

10WWW.SIGLENT.COM

SDS6000A Series Digital Storage Oscilloscope
Slope
Limit range

Rising, Falling<, >, in range, out of range

Time range

2 ns ～ 20 s, Resolution = 1 ns

Serial Trigger
Source

I2 C
SPI
UART
CAN
LIN

C1~C4/D0~D15
Standard: I2C, SPI, UART, CAN, LIN
Optional: CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT
Type: Start, Stop, Restart, No Ack, EEPROM, Address & Data, Data Length
Type: Data
Type: Start, Stop, Data, Parity Error
Type: All, Remote, ID, ID+Data, Error
Type: Break, Frame ID, ID+Data, Error

CAN FD（Optional）

Type: Start, Remote, ID, ID+Data, Error

FlexRay（Optional）

Type: TSS, Frame, Symbol, Errors

I2S（Optional）

Type: Data, Mute, Clip, Glitch, Rising Edge, Falling Edge

MIL-STD-1553B（Optional）

Type：Transfer, Word, Error, Timing

SENT（Optional）

Type: Start, Slow channel, Fast channel, Error

ProtocolSerial Decoder
Decoders

2

Threshold-4.1 ～ 4.1 div

List

1 ～ 7 lines

Decoder type
I2 C
SourceSignalAddress
SPI
SourceSignalEdge SelectChip selectBit Order
UART
SourceSignalData Width
Parity Check
Stop Bit
Idle Level
Bit Order
CAN
SourceLIN
LIN Version
SourceBaud Rate

Full duplex
C1~C4/D0~D15
SCL, SDA
7-bit, 10-bit
C1~C4/D0~D15
CLK, MISO, MOSI, CS
Rising, Falling
Active high, Active low, Clock timeout
LSB, MSB
C1~C4/D0~D15
RX, TX
5-bit, 6-bit, 7-bit, 8-bit
None, Odd, Even, Mark, Space
1-bit, 1.5-bit, 2-bit
Low, High
LSB, MSB
C1~C4/D0~D15
Ver 1.3, Ver 2.0
C1~C4/D0~D15
600 bps, 1200 bps, 2400 bps, 4800 bps, 9600 bps, 19200 bps, Custom

CAN FD（Optional）
SourceNominal Baud Rate
Data Baud Rate

C1~C4/D0~D15
10 kbps, 25 kbps, 50 kbps, 100 kbps, 250 kbps, 1 Mbps, Custom
500 kbps, 1 Mbps, 2 Mbps, 5 Mbps, 8 Mbps, 10 Mbps, Custom

FlexRay（Optional）
SourceBaud Rate

C1~C4/D0~D15
2.5 Mbps, 5 Mbps, 10 Mbps, Custom

I2S（Optional）
SourceSignalAudio Variant
WWW.SIGLENT.COM

C1~C4/D0~D15
BCLK, WS, DATA
Audio-I2S, Audio-LJ, Audio-RJ
11

SDS6000A Series Digital Storage Oscilloscope
Start Bits
Data Bits

0~31
1~32

MIL-STD-1553B（Optional）
SourceC1~C4

SENT（Optional）
SourceC1~C4/D0~D15

Manchester（Optional）
SourceBaud Rate

C1~C4
500 bps~5 Mbps

ARINC429（Optional）
SourceBaud Rate

C1~C4
12.5 kbps~100 kbps, tolerance 1%~20%

Word format

L/SDI/D/SSM，L/D/SSM，L/D

Measurement
Automatic Measurement
SourceC1~C4, D0~D15, Math, Ref, History, Zoom
Mode
Simple, Advanced
ScreenRange
Gated: inside screen, definable with separate Gate cursors
Custom Threshold
Upper, Middle, Lower
No. of Measurements
Display 12 measurements at the same time (Display mode = M2)
Max, Min, Pk-Pk, Top, Base, Amplitude, Mean, Cycle Mean, Stdev, Cycle Stdev, RMS, Cycle RMS,
Vertical Parameters
Median, Cycle Median, FOV, FPRE, ROV, RPRE, Level@Trigger
Period, Frequency, Time@max, Time@min, +Width, -Width, 10-90%Rise time, 90-10%Fall time, Rise
Horizontal Parameters
time, Fall time, +Burst Width, -Burst Width, +Duty Cycle, -Duty Cycle, Delay, Time@Middle, CycleCycle jitter+Area@DC, -Area@DC, Area@DC, Absolute Area@DC, +Area@AC, -Area@AC, Area@AC,
Miscellaneous
Absolute Area@AC, Cycles, Rising Edges, Falling Edges, Edges, Positive pulses, Negative pulses,
Parameters
Positive Slope, Negative Slope
Delay Parameters
Phase, FRFR, FRFF, FFFR, FFFF, FRLR, FRLF, FFLR, FFLF, Skew
Statistics
Current, Mean, Min, Max, Sdev, Count, Histogram, Trend, Track
Statistics Count
Unlimited, 1~1024
Statistics Count in one
Up to 65,000
frame
Cursors
SourceC1~C4、D0~D15、Math、Ref、Histogram

Type

Manual : Time X1, X2, (X1-X2), (1/ΔT); Vertical Y1, Y2, (Y1-Y2)
Track: Time X1, X2, (X1-X2)
Measure: indicates the measurement on specific parameter

Math
Trace
SourceOperation

FFT

F1, F2, F3, F4
C1~C4, Zoom, F1~F4
FFT, +, -, x, ÷, ∫dt, d/dt, √, Identity, Negation, |x|, Sign, ex, 10x, ln, lg, Interpolation, Max hold, Min
hold, ERES, Average, Filter, Formula Editor
Length: 8 Mpts, 4 Mpts, 2 Mpts, 1 Mpts, 512 kpts, 256 kpts, 128 kpts, 64 kpts, 32 kpts, 16 kpts, 8
kpts, 4 kpts, 2 kpts
Window: Rectangular, Blackman, Hanning, Hamming, Flattop
Display: Full Screen, Split, Exclusive
Mode: Normal, Max hold, Average
Tools: Peaks, Markers

Analysis
SearchSourceMode
Copy setting
Navigate
Type
Mask Test
Source12

C1~C4, History
Edge, Slope, Pulse, Interval, Runt
Copy from trigger, Copy to trigger
Search event, Time, History frame
C1~C4，Z1~Z4
WWW.SIGLENT.COM

SDS6000A Series Digital Storage Oscilloscope
Mask creating
Mask test speed
DVM
Source
Mode
Plot
Gate
Bode Plot
SourceSupported signal
sources
Sweep type

Auto (Create mask), Customized (Mask Editor)
Up to 18,000 frames/s
C1~C4
DC mean, DC RMS, AC RMS, Peak-peak, Amplitude
Bar, Histogram, Trend
20 ms

C1~C4
SAG1021I (Connection: USB),
SDG series waveform generators (Connection: USB, LAN)
Simple, Vari-level
Mode: Linear, Logarithmic
Frequency
Range: 10 Hz ~ 120 MHz
MeasureUpper cutoff frequency, Lower cutoff frequency, Bandwidth, Gain margin, Phase margin
Power Analysis (optional)
Power quality, Current Harmonics, Inrush current, Switching loss, Slew rate, Modulation, Output
Measureripple, Turn on/turn off, Transient response, PSRR, Efficiency, SOA
Histogram
SourceC1~C4
Type
Horizontal, Vertical, BothCounter
SourceC1~C4Frequency resolution
7 digits
Totalizer
Counter on edges, supports Gate and Trigger
Eye Diagram (optional)
SourceC1~C4
Clock recovery
Constant frequency, PLL
MeasureEye height, “1”level, “0”level, Eye amplitude, Eye width, Eye crossing, Average power, Q factor, TIE
Mask Test
Supported
Jitter Analysis (optional)
SourceC1~C4
Clock recovery
Constant frequency, PLL
Period, Frequency, +Width, -Width, +Duty cycle, -Duty cycle, Cycle-cycle jitter, Cycle-cycle +width,
MeasureCycle-cycle -Width, Cycle-cycle +Duty cycle, Cycle-cycle -Duty cycle, Bit Rate, Unit interval
TIE, RJ, DJ, DCD, DDJ, PJ, TJ@BER
Jitter decomposition
Statistics: Histogram, Track, Spectrum

Digital Channels
Max. Sampling Rate
Memory Depth
Min. Detectable Pulse
Width
Level Group
Level Range
Logic TypeSkew

1 GSa/s
50 Mpts/ch
3.3 ns
D0~D7, D8~D15-10 V~10 V
TTL, CMOS, LVCMOS3.3, LVCMOS2.5, Custom
D0~D15: ±1 sampling interval
Digital to Analog: ± (1 sampling interval +1 ns)

SAG1021I Waveform Generator (optional)
ChannelsMax. Output Frequency
Sampling Rate
Frequency Resolution
Frequency Accuracy
Vertical Resolution
Amplitude Range
Waveforms
Output ImpedanceProtection
Insulation Voltage
Sine
Frequency
WWW.SIGLENT.COM

1
25 MHz
125 MSa/s
1 μHz±50 ppm
14 bit-1.5 V ~ +1.5 V (into 50 Ω)-3 V ~ +3 V (into High-Z)
Sine, Square, Ramp, Pulse, DC, Noise, 45 Arbitrary
50 Ω ± 2%
Over voltage protection, Current limit±42 Vpk
1 μHz ～ 25 MHz
13

SDS6000A Series Digital Storage Oscilloscope
Offset accuracy (10kHz)
Amplitude flatness
SFDR
Harmonic distortion±(1%*offset setting value +3 mVpp)±0.3 dB, compare to 10 kHz, 2.5 Vpp into 50 Ω
DC ~ 1 MHz-60 dBc
1 MHz ~ 5 MHz -55 dBc
5 MHz ~ 25 MHz -50 dBc
DC ~ 5 MHz-50 dBc
5 MHz ~ 25 MHz-45 dBc

Square/Pulse
Frequency

1 μHz ～ 10 MHz

Duty cycle

1% ～ 99%

Edge< 24 ns (10% ～ 90%)

Overshoot< 3% (typical，1 kHz, 1 Vpp)

Pulse width
Jitter (cycle-cycle)
Ramp
Frequency> 50 ns< 500 ps + 10 ppm

Linearity
Channels< 0.1% of Pk-Pk (typical, 1 kHz, 1 Vpp, 50% symmetry)

1 μHz ～ 300 kHz
0% ～ 100%

DC
Offset range
Accuracy
Noise
Bandwidth (-3 dB)
Arb
Frequency
Waveform memory
Sample rate
Wave import±1.5 V (into 50 Ω)±3 V (into Hi-Z)±(|setting value|*1% + 3 mV)>25 MHz
1 μHz ～ 5 MHz
16 kpts
125 MSa/s
From EasyWaveX, from U-disk, directly from waveform data of analog channels

I/O
Front

Rear

USB 3.0 Host x2，
Calibration Signal: 1 kHz,3 V SquareUSB 2.0 Host x2,
USB 2.0 Device,
LAN: 10/100MbaseT (RJ45),
Micro SD Card,
External Trigger，EXT: ≤1.5 Vrms，EXT/5: ≤ 7.5Vrms,
Auxiliary Output: TRIG OUT(3.3 V LVCMOS), PASS/FAIL OUT(3.3 V TTL),
HDMI

Display
Display Type
Resolution
Contrast (typical)
Backlight (typical)
View angles (typical)

12.1 TFT LCD with capacitive touch screen
1280×800
1000:1450 nit
Top：85°, Bottom：85°, Left：85°, Right：85°

Display Setting
Range
Display Type
Persistence Time
Color Display
Language
Built-in Help System

8 x 10 gridDot, Vector
OFF, 0.1 s, 0.2 s, 0.5 s, 1 s, 5 s, 10 s, 30 s, infinite
Normal, Color; Supports customer trace color
Simplified Chinese, Traditional Chinese, English, French, Japanese, German, Spanish, Russian,
Italian, Portuguese
Simplified Chinese, English

Environmental
Temperature
14

Operating: 0 °C ~ 50 °C
Non-operating: -30 °C ~ 70 °C
WWW.SIGLENT.COM

SDS6000A Series Digital Storage Oscilloscope
Humidity
AltitudeElectromagnetic
Compatibility

SafetyRoHS

Operating: 5% ~ 90%RH, 30℃, degraded to 50%RH at 40 ℃
Non-operating: 5% ~ 95%
Operating: ≤ 3,048 m, 25 ℃
Non-operating: ≤12,192 m
Meets EMC directive (2014/30/EU), meets or exceeds IEC 61326-1:2012/EN61326-1:2013 (Basic)
CLASS A group 1
Conducted disturbance
CISPR 11/EN 55011
150 kHz-30 MHz
CLASS A group 1
Radiated disturbance
CISPR 11/EN 55011
30 MHz-1 GHz
Electrostatic discharge (ESD)
IEC 61000-4-2/EN 61000-4-2
4.0 kV (Contact),8.0 kV (Air)
10 V/m (80 MHz to 1 GHz);Radio-frequency
IEC 61000-4-3/EN 61000-4-3
3 V/m (1.4 GHz to 2 GHz);
electromagnetic field Immunity
1 V/m (2.0 GHz to 2.7GHz)
Electrical fast transients (EFT)IEC 61000-4-4/EN 61000-4-4
2kV (Input AC Power Ports)
1kV (Line to line)
SurgesIEC 61000-4-5/EN 61000-4-5
2kV (Line to ground)Radio-frequency continuous
IEC 61000-4-6/EN 61000-4-6
3 V, 0.15-80MHz
conducted Immunity
Voltage Dips:
0% UT during 1 cycle;
40% UT during 10/12 cycles;
Voltage dips and interruptions
IEC 61000-4-11/EN 61000-4-11
70% UT during 25/30 cycles
Voltage interruptions: 0% UTduring 250/300 cycles
UL 61010-1:2012/R: 2018-11; CAN/CSA-C22.2 No. 61010-1:2012/A1:2018-11.
UL 61010-2-030:2018; CAN/CSA-C22.2 No. 61010-2-030:2018.
EU 2015/863

Power Supply
Input Voltage & Frequency
Power consumption

100 ～ 240 Vrms 50/60Hz
100 ～ 120 Vrms 400 Hz
193 W max., 123 W typical, 4 W typical in standby mode

Mechanical
Dimensions

Length × Height × Width = 379mm × 288mm × 159mm

Weight

Net Weight 5.5 kg, Gross Weight 7.1 kg

WWW.SIGLENT.COM

15

SDS6000A Series Digital Storage Oscilloscope

Ordering Information
Model

Description

SDS6204A
SDS6104A
SDS6054A

2 GHz, 5 GSa/s, 4-CH, 500 Mpts/ch memory depth, 12.1” capacitive touch screen
1 GHz, 5 GSa/s, 4-CH, 500 Mpts/ch memory depth, 12.1” capacitive touch screen
500 MHz, 5 GSa/s, 4-CH, 500 Mpts/ch memory depth, 12.1” capacitive touch screenStandard Accessories

Quantity

USB cable
Quick start
Passive probe (SP3150A)
Certificate of calibration
Wireless mouse
Power cordProtective Cover

11
1/channel
11
11

Optional Accessories

Part No.

25 MHz isolated USB function/arbitrary waveform generator
16-channel logic probe
Power Analysis deskew fixture
STB3 demo signal source
USB-GPIB adapter
High-speed passive probe
High-speed active probe
High voltage probe
High-speed differential probe

Power rail probe
Rack Mount Kit
Bag

SAG1021I
SPL2016
DF2001A
STB3
USB-GPIB
SP6150A
SAP1000, SAP2500
HPB4010
SAP2500D
DPB1300/DPB4080/DPB5150/
DPB5150A/DPB5700/DPB5700A
CPL5100/CP4020/CP4050/CP4070/CP4070A
CP6030/CP6030A/CP6150/CP6500
SCP5030/SCP5030A/SCP5150/SCP5150A
SAP4000P
SDS6000-RMK
BAG-S2

Options

Part No.

Power Analysis (software)
Eye Diagram/Jitter Analysis (software)
I2S trigger & decode (software)
MIL-STD-1553B trigger & decode (software)
FlexRay trigger & decode (software)
CAN FD trigger & decode (software)
SENT trigger & decode (software)
Manchester decode (software)
ARINC429 decode (software)
500 MHz to 1 GHz bandwidth upgrade (software)
1 GHz to 2 GHz bandwidth upgrade (software)

SDS6000Pro-PA
SDS6000Pro-EJ
SDS6000Pro-I2S
SDS6000Pro-1553B
SDS6000Pro-FlexRay
SDS6000Pro-CANFDSDS6000Pro-SENT
SDS6000Pro-Manch
SDS6000Pro-ARINC
SDS6000-4BW10
SDS6000-4BW20

High voltage differential probe
Current probe

16

WWW.SIGLENT.COM

SDS6000A Series Digital Storage Oscilloscope

About SIGLENTSIGLENT is an international high-tech company, concentrating on R&D, sales,
production and services of electronic test & measurement instruments.

SIGLENT first began developing digital oscilloscopes independently in 2002.
After more than a decade of continuous development, SIGLENT has extended
its product line to include digital oscilloscopes, isolated handheld
oscilloscopes,

function/arbitrary

waveform

generators,

RF/MWsignalgenerators, spectrum analyzers, vector network analyzers, digital multimeters,
DC power supplies, electronic loads and other general purpose test
instrumentation. Since its first oscilloscope was launched in 2005, SIGLENT
has become the fastest growing manufacturer of digital oscilloscopes. We
firmly believe that today SIGLENT is the best value in electronic test &measurement.

Headquarters:
SIGLENT Technologies Co., Ltd
Add: Bldg No.4 & No.5, Antongda Industrial
Zone, 3rd Liuxian Road, Bao'an District,
Shenzhen, 518101, China
Tel: + 86 755 3688 7876
Fax: + 86 755 3359 1582
Email: sales@siglent.com
Website: int.siglent.com

North America:
SIGLENT Technologies America, Inc
6557 Cochran Rd Solon, Ohio 44139
Tel: 440-398-5800
Toll Free: 877-515-5551
Fax: 440-399-1211
Email: info@siglent.com

Follow us on
Facebook: SiglentTech

Website: www.siglentna.com

Europe:SIGLENT Technologies Germany GmbH
Add: Staetzlinger Str. 70
86165 Augsburg, Germany
Tel: +49(0)-821-666 0 111 0
Fax: +49(0)-821-666 0 111 22
Email: info-eu@siglent.com
Website: www.siglenteu.com
WWW.SIGLENT.COM

17