Quick Start Guide

PIPETBOY acu 2This quick guide is intended to provide a brief overview of the main features andoperation of PIPETBOY acu 2. For detailed information, please refer to the latest
Operating Instructions that can be found at www.integra-biosciences.com in several
languages.

Battery charge
Charge the Li-ion battery completely before first use.
Only use the original Li-ion battery for PIPETBOY acu 2(part no. 155 066)!
The NiMH battery for PIPETBOY acu is not compatible.
The supply voltage must meet the requirements of the mains
adapter: 100-240 VAC, 50/60 Hz.

Plug inLow battery

Charging battery

Charged battery~ 3.5 hRed blink

RedGreen

General speed setting

Plug in for turbo
speed

50 %
Speed
155921_V09_QSG_PIPETBOY_acu_2_EN.docx

100 %Speed

125 %Turbo speed
1/4Manual speed control
ASPI
RA
T
E

DI
S
PE
NS
E

Fast

Slow

Gravity dispense

Slow

Fast

Pump not running

Pump running

Pump running

Filter exchange
Press (1) and turn 1/8th counter clockwise (2)Replace the filter

If one side is colored, it must
face upwards
155921_V09_QSG_PIPETBOY_acu_2_EN.docx

2/4

GENERAL INFORMATIONPIPETBOY acu 2

Intended use
PIPETBOY acu 2 is a pipette controller designed for aspirating and dispensing aqueous solutions
with plastic or glass pipettes in a volume range of 1 to 100 ml. It is intended for measurement, controland laboratory use. Any use of this instrument in a medical or IVD setting is the responsibility of theuser.

Safety notes
1) Do not use or charge PIPETBOY acu 2 in an atmosphere with danger of explosion.
Do not pipette highly flammable liquids such as acetone or ether.
2) Comply with the material safety data sheet (MSDS) and with all safety guidelines when handling
dangerous substances. Use protective clothing and safety goggles. Never point a pipette inanyone’s direction.
3) Avoid pipetting of liquids whose vapours could attack the materials PA (polyamide), POM(polyoxymethylene), FPM (fluor-rubber), NBR (nitrile-rubber), CR (chloroprene) and silicone.
Corrosive vapours could also damage metallic parts inside the device.
4) Use an original INTEGRA Biosciences mains adapter only. Protect it from moisture otherwise
PIPETBOY acu 2 might be damaged.
Regardless of the listed safety notes, additionally applicable regulations and guidelines of tradeassociations, health authorities, trade supervisory offices, etc. must be observed.

Warranty
PIPETBOY pipette controllers are designed and manufactured following high standards in qualityand durability. They are warranted for a period of 12 months.
To benefit from a 3 year extended warranty, register yourPIPETBOY on www.pipetboy.info or www.integrabiosciences.com and click on Warranty registration.
Your warranty extension will be activated after online
submission of the registration form.

Equipment disposal
PIPETBOY acu 2 device must not be disposed of with unsorted municipal waste.
Dispose of PIPETBOY acu 2 and the battery separately in accordance with the laws
and regulations in your area governing disposal of devices.
In certain regions and countries, e.g. in EU member states, the distributor is obliged totake back this product free of charge at the end of life. Please contact your local
distributor for more details.

155921_V09_QSG_PIPETBOY_acu_2_EN.docx

3/4

Manufacturer and customer serviceINTEGRA Biosciences AG INTEGRA Biosciences Corp.
Tardisstrasse 201
CH-7205 Zizers, Switzerland

22 Friars Drive
Hudson, NH 03051, USAFind your local representative:
www.integra-biosciences.com
info@integra-biosciences.com

Direct sales country
China
Denmark, Sweden
FranceGermany, Austria
Japan
Switzerland
United Kingdom
United States, Canada

Telephone+86 21 5844 7203
+45 3173 5373
+33 1 34 30 76 76+49 6409 81999 15+81 3 5962 4936
+41 81 286 95 30+44 1635 797000+1 603 578 5800

Mail
info-cn@integra-biosciences.com
info-nordic@integra-biosciences.com
info-fr@integra-biosciences.com
info-de@integra-biosciences.com
info-jp@integra-biosciences.com
info-ch@integra-biosciences.com
info-uk@integra-biosciences.com
info-us@integra-biosciences.com

Declaration of conformity

INTEGRA Biosciences AG – 7205 Zizers, Switzerland
declares on its own responsibility that the devices
Description

ModelsPIPETBOY acu 2155000, 155015, 155016, 155017, 155018, 155019, 155022,155023, 155024

Accessories

153210, 153211, 153214, 153216, 155066

comply with:EU directives and regulations
2014/35/EU, 2014/30/EU, 2012/19/EC, 2011/65/EC, 2006/66/EC, 1907/2006, 2019/1782,1103/2010
GBR regulations
S.I. 2016/1101, S.I. 2016/1091, S.I. 2008/2164, S.I. 2013/3113, S.I. 2012/3032
Zizers, April 11, 2022

Urs Hartmann
CEODaniel Bächi
Head of Corporate Quality

For detailed CE declaration with regulations of other countries, please refer to the operating
instructions.

155921_V09_QSG_PIPETBOY_acu_2_EN.docx

4/4