％ ｃｈｅｃｋ

Super High Speed Measurement
High Stability Type
Up-to-date model correponding to High SpeedTaping MachineSuper High Speed Digital Resistance CheckerＭｄｅｌ
６０８Ｍｏｏｄｅｌ１
１６０８

RST ENGINEERING CO.,LTDKYOTO

JAPANＵｐ‑ｔｏ‑ｄａｔｅ ｍｏｄｅｌｃｏｒ
ｒｅｐｏｎｄｉｎｇｔｏＨｉｇｈｓｐｅｅｄＴａｐｉ
ｎｇＭａｃｈｉｎｅＳｕｐｅｒＨｉｇｈＳｐｅｅｄＤｉ
ｇｉｔ
ａｌＲｅｓｉｓｔａｎｃｅＣｈｅｃｋｅｒＭｏｄｅｌ１６０８
Contact check function☆ Measuring limit：１ｍΩ〜１００ＭΩ
Contact check function works super high speed at every measurement
is made after measurement and by 4 terminal measurement way, ☆ Display limit ： ±９．９９％ｏｒ±９９．９％
realized high level reliability.
(Contact error at contact resistance over 30Ω)☆ judgment value ：±０．００％〜±９．９９％
3 kinds of CE display function
setting limit(ＨＩ−ＱＵＡＬＩＴＹ ＯＮ)
When caused CE(contact error), CE point is displayed at TOLERANCE±００．０％〜±９９．９％
display position and can be confirmed CE point at a glance. Also,CE(ＨＩ−ＱＵＡＬＩＴＹ ＯＦＦ)point is printed at data of print.Same works at RS-232C* CE point at H terminal side ...... display as [CE4]☆ Measuring way ：4 terminals or 2term* CE point at L terminal side ........display as [CE8]-inals measurement* CE point at L&H terminal side....display as [CEC]● Standard functions : Standard functions : Function of 7000 memories at every measurements andstatistic analysis, print output

Ｓｐｅｃｉｆｉｃａｔｉｏｎ■ Measuring limit & Accuracy (value of accuracy are at 23 ℃ ±5 ℃)
ＳＴＡＮＤＡＲＤ
Measuring
Display limit
accuracysettig value
currency
HI-QUALITY ON
HI-QUALITY OFF
１０ ｍΩ 〜 ９９９ ｍΩ
５０ｍＡ
ｗｉｔｈｉｎα±３digit
ｗｉｔｈｉｎ±０.２％±１digit
１.００ Ω 〜 ９.９９ Ω２０ｍＡ±９．９９％
１０.０ Ω 〜 ９９.９ Ω
１０ｍＡ
HI-QUALITY
１００ Ω 〜 ９９９ Ω
５ｍＡONｗｉｔｈｉｎ±０．０３％±２ ｄｉｇｉｔ ｗｉｔｈｉｎ±０．１５％±１ ｄｉｇｉｔ
１.００ KΩ 〜 ９.９９ KΩ
５００μＡ
１０.０ KΩ 〜 ９９.９KΩ
５０μＡ±９９．９％
１００ KΩ 〜 １.００MΩ
５μＡ
HI-QUALITY
１．０１MΩ 〜 １０.０MΩ
０. ５μＡ
OFF±０.１％±１ｄｉｇｉｔ±０.２％±１ｄｉｇｉｔ
１０．１ MΩ 〜 １００ MΩ
０．０５μＡβ±１ｄｉｇｉｔ
β±１ｄｉｇｉｔ（ｍΩｃｈｅｃｋ）
２０ｍＡ
０〜９９９ｍΩ±０.２％±１ｄｉｇｉｔ
α：±（１０００/set value ｍΩ）×０．０１％
β：±（ set value ＭΩ/１００）％■ Measuring way Incase of OFF 2 & 4 terminal auto change
all ranges are 4 terminal■ Measuring time
STANDARD setting value
０１０ ｍΩ 〜 ９９９ ｍΩ
１.００ Ω 〜 １.００MΩ
１．０１ MΩ 〜 １０.０ MΩ
１０．１ MΩ 〜 １００ MΩ

In case of ON 2 & 4 terminal auto change
below 99.9kΩ(standard) .....4 terminals
over 100kΩ(standards)........ 2 terminals
For over 100kΩ, contact check works is cancelled

Remote startFree running
1.６ ｍｓｅｃ
６０ per/sec （６０Ｈｚ） ５０ per/sec （５０Ｈｚ）
1.４ ｍｓｅｃ〃５ ｍｓｅｃ〃
１９．５ｍｓｅｃ（６０Ｈｚ） ２２．８ｍｓｅｃ（５０Ｈｚ）
３０ per/sec （６０Ｈｚ） ２５回 per/sec （５０Ｈｚ）
The above times are the whole measured time from remote start input to measurment ending signal
output including contact check working■ Judgement value setting limit■ Input/Output signal■ Operation ambient■ Dimensions■ Power supply■ WeightUpper & Under limit±0.00% 〜 ± 9.99% [ HI-QUALITY ON ]±00.0% 〜 ± 99.9% [ HI-QUALITY OFF ]※ mΩ check000mΩ 〜 999mΩ
Input siganl = START, HOLD Output siganl = HI/GO/LO (judged result)
C.E. (Contact error)
E.O.C. ( Measurement ending))
Temp.: 0 ℃〜 50 ℃
Humidity : below 85%
328(W) × 99(H) × 300(D) mm
AC 100V/117V/220V/240V 50/60Hz
approx.30VA
approx.. 5.5 kg
ＯｐｔｉｏｎＲＳ−２３２Ｃ※ Subject to change specifications for improvement without notice.

RST ENGINEERING CO.,LTD.

HED OFFICE
NO.382, NISHIKANAGASAKI, KANSHUJI,
YAMASHINA-KU, KYOTO, 607-8221 JAPANＴＥＬ(０７５)５０１-５５０１ ＦＡＸ(０７５)５０１-７０９１
E-mail E-mail:info@rst-eng.co.jp
URL
http://www.rst-eng.co.jp/