SDS6000L
Low Profile
Digital Storage
Oscilloscope
Data SheetEN01B

SDS6208L / SDS6204L
SDS6108L / SDS6104L
SDS6058L / SDS6054L

Product Overview
SIGLENT’s SDS6000L series Low Profile
Digital Storage Oscilloscopes provide up to 8
analog channels + 16 digital channels in aspace-saving chassis size. This series isavailable in bandwidths of 2 GHz / 1 GHz / 500
MHz, feature a maximum sample rate of 5
GSa/s (10 GSa/s ESR) for each channel, anda maximum record length of 500 Mpts/ch.
The SDS6000L series employs Siglent’s SPO
technology with a maximum waveform
capture rate of up to 170,000 wfm/ s (normalmode, up to 750,000 wfm/s in Sequencemode), 256-level intensity grading display
function plus a color temperature display
mode. It also employs an innovative digitaltrigger system with high sensitivity and low
jitter. The trigger system supports multiple
powerful triggering modes including serial bus
triggering. Tools such as History waveformrecording, Search and Navigate functions,
Mask Test, Power Analysis and Eye/ Jitter
Analysis allow for extended waveform records
to be captured, stored, and analyzed. An
impressive array of measurement and mathcapabilities, options for a 25 MHz arbitrary
waveform generator, and serial decoding arealso features of the SDS6000L.
It can be used as a stand-alone oscilloscopeby being connected to an external display anda mouse. It can also be remotely controlled
over LAN thanks to the convenient built-in
web server. Combined with a 64-channel
synchronization distributor (SYN64), multiple
SDS6000L modules can be organized as ahigh-speed acquisition system with up to 512
channels. A complete SCPI command set
over the standard 1000M LAN connection
provides very fast data acquisition to speed
automated test applications.

S D S 6 0 00 L D a ta S h ee t

Key Features
8/4 analog channels + 1 external trigger. Designed
for expansion. Combine multiple units for a highspeed acquisition system with up to 512 channels.
Up to 2 GHz bandwidth with 5 GSa/s (10 GSa/s
ESR) sample rate at each channel
Low background noise, supports 0.5 mV/div to 10
V/div vertical scales
SPO technology
Waveform capture rates up to 170,000 wfm/s(normal mode), and 750,000 wfm/s(sequence mode)Supports 256-level intensity grading and color
temperature display modes
500 Mpts Record length in total for all 4
channels
Digital trigger system
Intelligent trigger: Edge, Slope, Pulse, Window,
Runt, Interval, Dropout, Pattern, Qualified, Nthedge, Setup / hold, Delay and Video (HDTV
supported). Zone Trigger simplifies advanced
triggering
Serial bus triggering and decoder, supports
protocols I2C, SPI, UART, CAN, LIN, CAN FD,
FlexRay, I2S, MIL-STD-1553B, SENT and
Manchester
Segmented acquisition (Sequence) mode, dividing
the maximum record length into multiple segments(up to 80,000), according to trigger conditions set
by the user, with a very small dead time between
segments to capture the qualifying event
History waveform record (History) function, themaximum recorded waveform length is 80,000
framesAutomatic measurements on 50+ parameters,
supports statistics with histogram, track, trend,
Gating measurement, and measurements on Math,History, Memory and Ref
4 Math traces (8 Mpts FFT, addition, subtraction,
multiplication, division, integration, differential,
square root, etc.), supports formula editor
Abundant data analysis functions such as Search,
Navigate, Digital Voltmeter, Counter, Waveform
Histogram, Power Analysis and Eye/Jitter Analysis
16 digital channels
25 MHz function / arbitrary waveform generator,
built-in multiple predefined waveforms
Interfaces include: 4x USB Hosts, USB Device(USBTMC), 1000M LAN（VXI-11/ Telnet/ Socket）,
micro SD card, Pass/Fail, Trigger Out, HDMI,
10MHz In、10MHz Out
Built-in web server supports remote control over
the LAN port using a web browser. Supports SCPI
remote control commands. Supports externalmouse and keyboard.

1Models and Key Specifications
Model
Channel
8
4

Model

2 GHz

1 GHz

500 MHz

SDS6208L
SDS6204L

SDS6108L
SDS6104L

SDS6058L
SDS6054L

SDS6208L
SDS6204L

SDS6108L
SDS6104L

SDS6058L
SDS6054L

Channel

8/4 + EXT

Bandwidth

2 GHz

Sample rate
(Max.)

5 GSa/s (10 GSa/s ESR) @ each channel

Memory depth(Max.)

500 Mpts/ch (single-channel)
250 Mpts/ch (dual-channel)
125 Mpts/ch (3 or 4 channels)

Waveform
capture rate
(Max.)
Verticalresolution

1 GHz

500 MHz

Normal mode: 170,000 wfm/s;
Sequence mode: 750,000 wfm/s

Serial trigger
and decode

8-bit
Up to 12-bit in ERES mode, equivalent to 16-bit Hi-Res mode
Edge, Slope, Pulse width, Window, Runt, Interval, Dropout, Pattern, Video,
Qualified, Nth edge, Setup/hold, Delay, Serial
Standard: I2C, SPI, UART, CAN, LIN
Optional: CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT, Manchester (decode only)

Measurement

50+ parameters, statistics, histogram, trend, and tracking supportedMath

4 traces
8 Mpts FFT, +, -, x, ÷, ∫dt, d/dt, √ , Identity, Negation, Absolute, Sign, ex, 10x, ln, lg,
Interpolation, MaxHold, MinHold, ERES, Average. Supports formula editor

Data analysis

Search, Navigate, History, Mask Test, Digital Voltmeter, Counter, Waveform
Histogram, Power Analysis, Eye / Jitter Analysis

Digital channel

16-channel; maximum sample rate up to 1 GSa/s; record length up to 50 Mpts

Waveform
generator(optional)

Built-in single-channel waveform generator, frequency up to 25 MHz, 125 MSa/s
sample rate, 16 kpts waveform memory

I/O

HDMI (1280*800), USB 3.0 Host x2, USB 2.0 Host x2, USB 2.0 Device (USBTMC),
1000M LAN (SCPI over VXI-11, SCPI over Telnet (port:5024), SCPI over Socket(port:5025), micro SD card, External trigger, Auxiliary output (TRIG OUT,PASS /FAIL), 10 MHz In, 10 MHz Out

Probe(Standard)

1x 500 MHz passive probe supplied for each channel

Trigger typeMultiple-channel Acquisition System

2

ChannelUp to 512

JitterWithin a unit: < 100 ps,rms

Skew

Without skew calibration, within a unit: < 100 ps
With skew calibration, within a unit: < 100 ps

Between units:< 250 ps,rms
Between units: < 500 ps
Between units: < 150 ps

SDS6000L Data SheetFunctions & Characteristics
Up-to 8 analog channels

Monitor and control up-to 8 analog channels and 16 logic channels with one SDS6000L mainframe. Easily
view all data using the convenient VNC web control

S D S 6 0 00 L D a ta S h ee t

3Flexible Multi-channel High-speed Acquisition System

Connection
Diagram

Sync

Host
LAN connectiion
Synchronization signal
10MHz In (optional)

SYN64

Oscilloscope #1

Oscilloscope #2

Network Switch

Oscilloscope #3

10MHz Out (optional)

Oscilloscope #NStandard sizes: 4 - channel models – 1U,8-channel models – 2U

Multiple units are combined to create a high-speed acquisition system with up to 512 channels by
being triggered with low-skew synchronization signals from the 64-channel synchronization
distributor SYN64

The host can access each unit over 1000M LAN. A complete SCPI command set as well as
LabVIEW and IVI drivers are provided for easy data acquisition. The LAN port is LXI compliant.



Sample clocks are synchronized between all units in the test system by cascading the 10 MHz In
and 10 MHz Out clocks in a daisy chain

4

SDS6000L Data Sheet

High Waveform Update Rate

Deep Record LengthWith a waveform update rate of up to 170,000
wfm/s, the oscilloscope can easily capture unusual
or low-probability events. In Sequence mode, thewaveform capture rate can reach 750,000 wfm/s.

Using a hardware-based Zoom technique andrecord length of up to 500 Mpts, users can select aslower timebase without compromising the sample
rate, and then quickly zoom in to focus on the areaof interest.

Multiple Trigger Functions

Trigger Zone

Edge, Slope, Pulse, Video, Windows, Runt,
Interval, Dropout, Pattern, Qualified, Nth edge,
Setup/ hold, Delay, and serial trigger.

Trigger Zone is available for advanced triggering.

Advanced Math Function

In addition to the traditional (+, -, X, /) operations,
FFT, integration, differential, square root, and moreare supported. Formula Editor is available for more
complex operations. 4 math traces are available.

S D S 6 0 00 L D a ta S h ee t

Hardware-accelerated FFT supports up to 8 Mpts
operation. This provides high-frequency resolution
with a fast refresh rate. The FFT function also
supports a variety of window functions so that it canadapt to different spectrum measurement needs.
Three modes (Normal, Average, and Max hold) cansatisfy different requirements for observing thepower spectrum. Auto peak detection and markers
are supported.

5

Measurements of a Variety of Parameters

Parameter Statistics Function

Parameter measurements include 4 categories:
horizontal, vertical, miscellaneous, and CH delay
providing a total of 50+ different types ofmeasurements. Measurements can be performed
within a specified gate period. Measurements on
Math, Reference, and History frames aresupported.

Statistics show the current value, maximum value,
minimum value, standard deviation, and meanvalue of up to 12 parameters simultaneously. A
histogram is available to show the probability
distribution of a parameter. Trend and Track areavailable to show the parameter value vs. time.
For horizontal parameters such as period, all results
are extracted from a frame, instead of justcalculating the first one. This accelerates statistics
on horizontal measurements much more andenables distribution observation in a frame using
Histogram and Track.

History Mode

Sequence Mode

The history function can record up to 80,000 framesof waveforms. The recording is executed
automatically so that the customer can playback thehistory waveforms at any time to observe unusual
events and quickly locate the area of interest using
cursors or measurements. The failed frames of the
Mask Test can be stored as history frames.

Segmented memory collection will store thewaveform into multiple memory segments (up to
80,000) and each segment will store a triggered
waveform as well as the dead time information. Theinterval between segments can be as small as
1.3μs. All of the segments can be played back using
the History function.

6

SDS6000L Data SheetSearch and Navigate

Serial Bus Decode

The oscilloscope can search events specified by
the user in a frame. Events flagged by the Search
can be recalled automatically using Navigate. It canalso navigate by time (delay position) and history
frames.

Display the decoded characters through the events
list. Bus protocol information can be quickly andintuitively displayed in tabular form. I2C, SPI, UART,
CAN, LIN, CAN FD, FlexRay, I2S, MIL-STD-1553B,
SENT, and Manchester are supported.

Hardware-based High Speed Mask Test
Function

The oscilloscope utilizes a hardware-based Mask
Test function, performing up to 18,000 Pass / Faildecisions each second. It is easy to generate userdefined test templates to provide trace mask
comparisons, making it suitable for long-term signal
monitoring or automated production line testing.

S D S 6 0 00 L D a ta S h ee t

Built-in Mask Editor application helps to createcustom masks.

7

Eye / Jitter Analysis

Supports eye diagram and jitter analysis / measurement. It can automatically extract the embedded
reference clock from serial data and create an eye diagram. Measurement on multiple eye / jitterparameters is provided and mask testing of eye diagrams is supported.
Power Analysis (Optional)
The Power Analysis option provides a full suite of
power measurements and analysis, which greatly
improves the measurement efficiency in switching
power supplies and power devices design.

Digital Channels / MSO

25 MHz Function/Arbitrary Waveform
Generator (Optional)

Four analog channels plus 16 digital channels
enable users to acquire and trigger the waveforms
and then analyze the pattern, simultaneously with
one instrument.

The built-in Function / Arbitrary waveform generator
can output waveform with up to 25MHz frequency
and ±3V amplitude. Six basic waveforms plus
multiple types of arbitrary waveforms are built-in.

8

SDS6000L Data Sheet

Complete Connectivity

HDMI (1280 x 800) x1, USB Host 3.0 x2, USB Host 2.0 x2, USB Device 2.0 (USBTMC) x1, 1000M LAN(VXI-11 / Telnet / Socket)x1, micro SD card x1, Auxiliary output (Pass / Fail, Trigger Out) x1, 10 MHz In x1,
10 MHz Out x1

S D S 6 0 00 L D a ta S h ee t

9Specifications
All specifications are not guaranteed unless the following conditions are met:

The oscilloscope calibration period is current

The oscilloscope has been working continuously for at least 30 minutes at the specified temperature(18℃ ~ 28℃)
Acquire (analog)
Sample rate
Memorydepth *2,3
Waveform
update rate

5 GSa/s (10 GSa/s (ESR*1) @ each channel
500 Mpts/ch (single-channel)
250 Mpts/ch (dual-channel)
125 Mpts/ch (3 or 4 channels)
Normal mode: up to 170,000 wfm/s
Sequence mode: up to 750,000 wfm/s

Intensity
grading

256-level

Peak detect

200 psAverage

4, 16, 32, 64, 128, 256, 512, 1024, 2048, 4096, 8192

ERES

Enhanced bit: 0.5, 1, 1.5, 2, 2.5, 3, 3.5, 4 bit

Sequence

Up to 80,000 segments, interval between triggers = 1.3 μs min.

History

Up to 80,000 frames

Interpolation

sinx/x, x* 1: ESR: Enhanced Sample Rate provides better measurement accuracy by using 2x interpolation* 2: In Average and ERES modes, the memory depth is 25 Mpts/ch* 3: C1~C4 as a group and C5~C8 as another group. In a group:
Single-channel: Only one channel is enabled in a group
Dual-channel: Two channels are enabled in a group

Vertical (analog)

SDS6208L
SDS6204L

SDS6108L
SDS6104L

SDS6058L
SDS6054L

Channel

8/4 + EXT

Resolution

8-bit
Up to 12-bit in ERES mode, equivalent to 16-bit Hi-Res mode

Bandwidth(-3dB)@50Ω

2 GHz *1

1 GHz

500 MHz230 ps350 ps550 psRise time@50Ω
(typical)Bandwidth(-3dB)@1MΩ, withprobe
Bandwidth in ERES
mode (typical)

10500 MHz
0.5-bit: 0.25*Sample rate, up to the analog bandwidth
1-bit: 0.115*Sample rate, up to 1.15 GHz, limited by the analog bandwidth
1.5-bit: 0.055*Sample rate, up to 550 MHz, limited by the analog bandwidth
2-bit: 0.028*Sample rate, up to 280 MHz
2.5-bit: 0.014*Sample rate, up to 140 MHz
3-bit: 0.007*Sample rate, up to 70 MHz
SDS6000L Data Sheet

Range3.5-bit: 0.0035*Sample rate, up to 35 MHz
4-bit: 0.0017*Sample rate, up to 17 MHz
8 divisions

Vertical scale(probe 1X)

1 MΩ: 0.5 mV/div – 10 V/div
50 Ω: 0.5 mV/div – 1 V/div

DC gain accuracy± 1.5%

Offset accuracy± (1% of the offset setting + 0.5% of full scale + 0.02% of max offset + 1mV)

Offset range(probe 1X)

1MΩ: 0.5 mV/div ~ 5 mV/div: ± 1.6 V
5.1 mV/div ~ 10 mV/div: ± 4 V
10.2 mV/div ~ 20 mV/div: ± 8 V
20.5 mV/div ~ 100 mV/div: ± 16 V
102 mV/div ~ 200 mV/div: ± 80V
205 mV/div ~ 1 V/div: ± 160 V
1.02 V/div ~ 10 V/div: ± 400 V
50 Ω: 0.5 mV/div ~ 5 mV/div: ± 1.6 V
5.1 mV/div ~ 10 mV/div: ± 4 V
10.2 mV/div ~ 20 mV/div: ± 8 V
20.5 mV/div ~ 1 V/div: ± 10 V

Bandwidth limit

Hardware Bandwidth limit: 20 MHz, 200 MHz

Low-frequency
response(AC coupling -3 dB)

6 Hz (typical)

Overshoot(100 mV/div, 150 psedge @50 Ω, typical)

15%

Coupling

DC, AC, GND

Impedance(1 MΩ ± 2%) || (20 pF ± 3pF)
50 Ω: 50 Ω ± 2%

Max. Input voltage

1 MΩ < 400 Vpk(DC + AC), DC~10 kHz
50 Ω < 5 Vrms, ±10V Peak

SFDR

> 45 dBc

CH to CH Isolation(@50Ω)

70 dB up to 200 MHz
60 dB up to 500 MHz
50 dB up to 1 GHz
40 dB up to 2 GHz

Probe Attenuation

1X, 10X, 100X, custom

10%

5%

* 1: The bandwidth is 1 GHz below 2.3 mV/div

SDS6208L
SDS6204L

Horizontal
Time scale

0.1 ns/div – 1000 s/div

Range

10 divisions

Display mode

Y-T, X-Y, Roll

SD S6000L Data She et

SDS6108L
SDS6104L
0.2 ns/div – 1000 s/div

SDS6058L
SDS6054L
0.5 ns/div – 1000 s/div

11

Roll mode

≥ 50 ms/div

Skew(CH1~CH8)< 100 psTime base
Accuracy± 2 ppm initial (0~50℃); ± 0.5 ppm 1st year aging; ± 3 ppm 20-year aging

12

SDS6000L Data SheetTrigger
Mode

Auto, Normal, Single

Level

Internal: ± 4.5 div from the center of the screen
EXT: ± 0.61 V
EXT/5: ± 3.05 V

Hold off range

By time: 8 ns ~ 30 s (8 ns step)
By event: 1 ~ 108

Coupling

CH1~CH8
DC: Passes all components of the signal
AC: Blocks DC components and attenuates signals below 15 Hz
LFRJ: Attenuates the frequency components below 2.4 MHz
HFRJ: Attenuates the frequency components above 1.3 MHz
Noise RJ: Increases the trigger hysteresis
EXT
DC: Passes all components of the signal
AC: Blocks DC components and attenuates signals below 15 Hz
LFRJ: Attenuates the frequency components below 2.5 MHz
HFRJ: Attenuates the frequency components above 1.3 MHz

Accuracy (typical)

CH1 ~ CH8: ± 0.2 div
EXT: ± 0.3 div

CH1 ～ CH8:
Sensitivity>10 mV/div:

Noise RJ = OFF± 0.26 div

Noise RJ = ON
± 0.33 div

5 mV/div~10 mV/div:± 0.26 div± 0.33 div≤ 2 mV/div:± 0.5 div

± 0.5 div

EXT:

200 mVpp, DC ~ 10 MHz
300 mVpp, 10 MHz ~ bandwidth (300 MHz)

EXT/5:

1 Vpp, DC ~ 10 MHz
1.5 Vpp, 10 MHz ~ bandwidth (300 MHz)

Jitter

CH1 ~ CH8:
< 9 ps RMS (typical) for ≥ 300 MHz sine and ≥ 6 divisions peak to peak amplitude
for vertical gain settings from 2.5 mV/div to 10 V/div< 5 ps RMS (typical) for ≥ 500 MHz sine and ≥ 6 divisions peak to peak amplitude
for vertical gain settings from 2.5 mV/div to 10 V/div
EXT: < 50 ps rms

Displacement

Pre-Trigger: 0 ~ 100% memory
Delay-Trigger: 0 ~ 10,000 div

Zone

Up to 2 zones
Source: CH1~CH8
Property: Intersect, Not Intersect

Edge Trigger
SourceCH1~CH8 / EXT / (EXT/5) / AC Line / D0~D15

Slope

Rising, Falling, Rising & Falling

Slope Trigger
SourceCH1 ~ CH8

Slope

Rising, Falling

SDS6 000L Data She et

13

Limit range<, >, in range, out of range

Time range

2 ns ~ 20 s, Resolution = 1 ns

Pulse Width Trigger
SourceCH1~CH8 / D0~D15

Polarity+wid, -wid

Limit range<, >, in range, out of range

Time range

2 ns ~ 20 s, Resolution = 1 ns

Video Trigger
SourceCH1 ~ CH8Standard

NTSC, PAL, 720p/50, 720p/60, 1080p/50, 1080p/60, 1080i/50, 1080i/60, Custom

Synchronization

Any, SelectTrigger Condition

Line, Field

Window Trigger
SourceCH1 ~ CH8

Window type

Absolute, Relative

Interval Trigger
SourceCH1 ~ CH8 / D0~D15

Slope

Rising, Falling

Limit range<, >, in range, out of range

Time range

2 ns ~ 20 s, Resolution = 1 ns

Dropout Trigger
SourceCH1 ~ CH8 / D0~D15

Timeout type

Edge, State

Slope

Rising, Falling

Time range

2 ns ~ 20 s, Resolution = 1 ns

Runt Trigger
SourceCH1 ~ CH8

Polarity

Positive, Negative

Limit range<, >, in range, out of range

Time range

2 ns ~ 20 s, Resolution = 1 ns

Pattern Trigger
SourceCH1 ~ CH8 / D0 ~ D15

Pattern Setting

Don’t Care, Low, High

Logic

AND, OR, NAND, NOR

Limit range<, >, in range, out of range

Time range

2 ns ~ 20 s, Resolution = 1 ns

Qualified Trigger
TypeState, State with Delay, Edge, Edge with Delay

Qualified Source

CH1~CH8 / D0~D15

14

SDS6000L Data SheetEdge Trigger
SourceCH1~CH8 / D0~D15

Nth Edge Trigger
SourceCH1 ~ CH8 / D0 ~ D15

Slope

Rising, Falling

Idle time

8 ns ~ 20 s, Resolution = 1 ns

Edge Number

1 ~ 65535

Delay Trigger
Source A

CH1 ~ CH8 / D0~D15

Source B

CH1 ~ CH8 / D0~D15

Slope

Rising, Falling

Limit range<, >, in range, out of range

Time range

2 ns ~ 20 s, Resolution = 1 ns

Serial Trigger
SourceCH1~CH8 / D0~D15

ProtocolStandard: I2C, SPI, UART, CAN, LIN
Optional: CAN FD, FlexRay, I2S, MIL-STD-1553B, SENT

I2C

Type: Start, Stop, Restart, No Ack, EEPROM, Address & Data, Data Length

SPI

Type: Data

UART

Type: Start, Stop, Data, Parity Error

CAN

Type: All, Remote, ID, ID+Data, Error

LIN

Type: Break, Frame ID, ID+Data, Error

CAN FD(Optional)

Type: Start, Remote, ID, ID+Data, Error

FlexRay(Optional)

Type: TSS, Frame, Symbol, Errors

I2S (Optional)

Type: Data, Mute, Clip, Glitch, Rising Edge, Falling Edge

MIL-STD-1553B(Optional)

Type: Transfer, Word, Error, Timing

SENT (Optional)

Type: Start, Slow channel, Fast channel, Error

Serial Decoder
Decoders

2

Threshold- 4.1 ~ 4.1 div

List

1 ~ 7 lines

Decoder type

Full duplex

I2C
SourceCH1 ~ CH8 / D0 ~ D15

SignalSCL, SDA

Address

7-bit, 10-bit

SPI

SDS6 000L Data She et15

Source

CH1 ~ CH8 / D0 ~ D15

Signal

CLK, MISO, MOSI, CS

Edge SelectRising, Falling

Chip selectActive high, Active low, Clock timeout

Bit Order

LSB, MSB

UART
SourceCH1 ~ CH8 / D0 ~ D15

SignalRX, TX

Data Width

5-bit, 6-bit, 7-bit, 8-bit

Parity Check

None, Odd, Even, Mark, Space

Stop Bit

1-bit, 1.5-bit, 2-bit

Idle Level

Low, HighBit Order

LSB, MSB

CAN
SourceCH1 ~ CH8 / D0 ~ D15

LIN
LIN Version

Ver 1.3, Ver 2.0

SourceCH1 ~ CH8 / D0 ~ D15

Baud Rate

600 bps, 1200 bps, 2400 bps, 4800 bps, 9600 bps, 19200 bps, Custom

CAN FD（Optional）
SourceCH1 ~ CH8 / D0 ~ D15

Nominal BaudRate

10 kbps, 25 kbps, 50 kbps, 100 kbps, 250 kbps, 1 Mbps, Custom

Data Baud Rate

500 kbps, 1 Mbps, 2 Mbps, 5 Mbps, 8 Mbps, 10 Mbps, Custom

FlexRay（Optional）
SourceCH1 ~ CH8 / D0 ~ D15

Baud Rate

2.5 Mbps, 5 Mbps, 10 Mbps, Custom

I2S（Optional）
SourceCH1 ~ CH8 / D0 ~ D15

Signal

BCLK, WS, DATA

Audio Variant

Audio-I2S, Audio-LJ, Audio-RJ

Start Bits

0 ~ 31

Data Bits

1 ~ 32MIL-STD-1553B（Optional）
SourceCH1 ~ CH8SENT（Optional）
SourceCH1 ~ CH8 / D0 ~ D15

Manchester（Optional）
SourceCH1 ~ CH8

Baud Rate

500 bps ~ 5 Mbps

16

SDS6000L Data SheetMeasurement
Automatic Measurement
SourceCH1~CH8,D0~D15, Z1~Z8, F1~F4, M1~M4, Ref, History

Mode

Simple, Advanced

Range

ScreenGated: inside screen, definable with separate Gate cursors

CustomThreshold

Upper, Middle, Lower

No. ofMeasurements

Display 12 measurements at the same time (Display mode = M2)

VerticalParameters

Max, Min, Pk-Pk, Top, Base, Amplitude, Mean, Cycle Mean, Stdev, Cycle Stdev,
RMS, Cycle RMS, Median, Cycle Median, FOV, FPRE, ROV, RPRE,
Level@Trigger

Horizontal
Parameters

Period, Frequency, Time@max, Time@min, +Width, -Width, 10-90%Rise time,
90-10%Fall time, Rise time, Fall time, +Burst Width, -Burst Width, +Duty Cycle,-Duty Cycle, Delay, Time@Middle, Cycle-Cycle jitter

Miscellaneous
Parameters+Area@DC, -Area@DC, Area@DC, Absolute Area@DC, +Area@AC, -Area@AC,
Area@AC, Absolute Area@AC, Cycles, Rising Edges, Falling Edges, Edges,
Positive pulses, Negative pulses, Positive Slope, Negative Slope

Delay Parameters

Phase, FRFR, FRFF, FFFR, FFFF, FRLR, FRLF, FFLR, FFLF, Skew, Tsu@R,
Tsu@F, Th@R, Th@F

Statistics

Current, Mean, Min, Max, Sdev, Count, Histogram, Trend, Track

Statistics Count

Unlimited, 1~1024

Cursors
SourceCH1~CH8, D0~D15, F1~F4, M1~M4, Ref, Histogram

Type

Manual: Time X1, X2, (X1-X2), (1/ΔT); Vertical Y1, Y2, (Y1-Y2)
Track: Time X1, X2, (X1-X2)
Measure: Indicates the measurement of a specific parameter

Math
Trace

F1, F2, F3, F4

SourceCH1~CH8, Z1~Z8, F1~F4, M1~M4

Operation

FFT, +, -, x, ÷, ∫dt, d/dt, √, Identity, Negation, |x|, Sign, ex, 10x, ln, lg,
Interpolation, Max hold, Min hold, ERES, Average, Formula Editor

FFT

Length: 8 Mpts, 4 Mpts, 2 Mpts, 1 Mpts, 512 kpts, 256 kpts, 128 kpts, 64 kpts,
32 kpts, 16 kpts, 8 kpts, 4 kpts, 2 kpts
Window: Rectangular, Blackman, Hanning, Hamming, Flattop
Display: Full Screen, Split, Exclusive
Mode: Normal, Max hold, Average
Tools: Peaks, Markers

SDS6 000L Data She et17Analysis
SearchSourceCH1 ~ CH8, History

Mode

Edge, Slope, Pulse, Interval, Runt

Copy setting

Copy from trigger, Copy to trigger

Navigate
Type

Search event, Time, History frame

Mask Test
SourceCH1 ~ CH8，Z1 ~ Z8

Mask creating

Auto (Create mask), Customized (Mask Editor)

Mask test speed

Up to 18,000 frames/s

DVM
SourceCH1 ~ CH8

Mode

DC mean, DC RMS, AC RMS, Peak-peak, Amplitude

Plot

Bar, Histogram, Trend

Gate

20 ms

Bode Plot
SourceCH1 ~ CH4

Supported signal
sources

SAG1021I (Connection: USB),
SDG series waveform generators (Connection: USB, LAN)Power Analysis (optional)
SourceCH1 ~ CH4

MeasurePower quality, Current Harmonics, Inrush current, Switching loss, Slew rate,
Modulation, Output ripple, Turn on/turn off, Transient response, PSRR, Efficiency,
SOA

Histogram
SourceCH1 ~ CH8

Type

Horizontal, Vertical, BothCounter
SourceFrequency
resolution
Totalizer

CH1 ~ CH8
7 digits
Counter on edges, supports Gate and Trigger

Eye Diagram (optional)
SourceCH1 ~ CH8

Clock recovery

Constant frequency, PLL

MeasureEye height, “1” level, “0” level, Eye amplitude, Eye width, Eye crossing, Average
power, Q factor, TIE

Mask Test

Supported

Jitter Analysis (optional)
Source18

CH1 ~ CH8
SDS6000L Data SheetClock recovery

Constant frequency, PLL

MeasurePeriod, Frequency, +Width, -Width, +Duty cycle, -Duty cycle, Cycle-cycle jitter,
Cycle-cycle +width, Cycle-cycle -Width, Cycle-cycle +Duty cycle, Cycle-cycle-Duty cycle, Bit Rate, Unit interval

Jitterdecomposition

TIE, RJ, DJ, DCD, DDJ, PJ, TJ@BER
Statistics: Histogram, Track, Spectrum

Digital Channels
Max. Sampling
Rate
Memory Depth
Min. Detectable
Pulse Width
Level Group

1 GSa/s
50 Mpts/ch
3.3 ns
D0~D7, D8~D15

Level Range-10 V~10 V

Logic Type

TTL, CMOS, LVCMOS3.3, LVCMOS2.5, Custom

Skew

D0~D15: ±1 sampling interval
Digital to Analog: ± (1 sampling interval +1 ns)

Waveform Generator (optional)
Channels1Max. Output
Frequency

25 MHzSampling Rate

125 MSa/s

Frequency
Resolution

1 μHz

Frequency
Accuracy
VerticalResolution
Amplitude Range
Waveforms
OutputImpedance
Protection± 50 ppm
14-bit-1.5 V ~ +1.5 V (into 50 Ω)-3 V ~ +3 V (into High-Z)
Sine, Square, Ramp, Pulse, DC, Noise, 45 Arbitrary
50 Ω ± 2%
Over-voltage protection, Current limit

Sine
Frequency

1 μHz ~ 25 MHz

Offset accuracy(10 kHz)± (1%*offset setting value +3 mVpp)

Amplitude
flatness± 0.3 dB, compared to 10 kHz, 2.5 Vpp into 50 Ω

SFDR

DC ~ 1 MHz-60 dBc
1 MHz ~ 5 MHz -55 dBc
5 MHz ~ 25 MHz -50 dBc

SDS6 000L Data She et

19Harmonic
distortion

DC ~ 5 MHz-50 dBc
5 MHz ~ 25 MHz -45 dBc

Square / Pulse
Frequency

1 μHz ~ 10 MHz

Duty cycle

1% ~ 99%

Edge< 24 ns (10% ~ 90%)

Overshoot< 3% (typical，1 kHz, 1 Vpp)

Pulse width> 50 ns

Jitter (cycle-cycle)< 500 ps + 10 ppm

Ramp
Frequency

1 μHz ~ 300 kHz

Linearity< 0.1% of Pk-Pk (typical, 1 kHz, 1 Vpp, 50% symmetry)

Channels

0% ~ 100%

DC
Offset range± 1.5 V (into 50 Ω)± 3 V (into Hi-Z)

Accuracy± (|setting value|*1% + 3 mV)

NoiseBandwidth (-3 dB)>25 MHzArb
Frequency
Waveform
memorySample rate

1 μHz ～ 5 MHz

Wave import

From EasyWaveX, from U-disk, directly from waveform data of analog channels

16 kpts
125 MSa/s

I/O
Front

USB 3.0 Host x2，
Calibration Signal: 1 kHz,3 V Square

Rear

USB 2.0 Host x2,
USB 2.0 Device,
LAN: 10 / 100M / 1000M BaseT (RJ45),
Micro SD Card,
External Trigger, EXT: ≤1.5 Vrms, EXT/5: ≤ 7.5Vrms,
Auxiliary Output: TRIG OUT(3.3 V LVCMOS), PASS/FAIL OUT(3.3 V TTL),
HDMI
10 MHz In, 10 MHz Out
AWG

20

SDS6000L Data SheetDisplay SettingRange

8 x 10 grid

Display Type

Dot, Vector

Persistence TimeOFF, 0.1 s, 0.2 s, 0.5 s, 1 s, 5 s, 10 s, 30 s, infinite

Color Display

Normal, Color; Supports customer trace color

Language

Simplified Chinese, Traditional Chinese, English, French, Japanese, German,
Spanish, Russian, Italian, Portuguese

Built-in Help
System

Simplified Chinese, English

Environmental
Temperature

Operating: 0℃ ~ 50℃
Non-operating: -30℃ ~ 70℃

Humidity

Operating: 5% ~ 90%RH, 30℃, degraded to 50%RH at 40℃
Non-operating: 5% ~ 95%

AltitudeOperating: ≤ 3,048 m, 25℃
Non-operating: ≤12,192 m
Meets EMC directive (2014/30/EU),meets or exceeds IEC 61326-1:2012/EN61326-1:2013 (Basic)

Electromagnetic
Compatibility

Conducted disturbance

CISPR 11 / EN
55011

CLASS A group 1
150 kHz - 30 MHz

Radiated disturbance

CISPR 11 / EN
55011

CLASS A group 1
30 MHz - 1 GHz

Electrostatic discharge(ESD)

IEC 61000-4-2 / EN
61000-4-2

4.0 kV (Contact),
8.0 kV (Air)Radio-frequency
electromagnetic field
Immunity

IEC 61000-4-3 / EN
61000-4-3

10 V/m (80 MHz to 1 GHz);
3 V/m (1.4 GHz to 2 GHz);
1 V/m (2.0 GHz to 2.7GHz)

Electrical fast transients(EFT)

IEC 61000-4-4 / EN
61000-4-4

2kV(Input AC Power Ports)

SurgesIEC 61000-4-5 / EN
61000-4-5

1kV (Line to line)
2kV (Line to ground)Radio-frequency
continuous conducted
Immunity

IEC 61000-4-6 / EN
61000-4-6

3 V, 0.15-80MHz

IEC 61000-4-11/EN
61000-4-11

Voltage Dips:
0% UT during 1 cycle;
40% UT during 10/12 cycles;
70% UT during 25/30 cycles
Voltage interruptions: 0% UTduring 250/300 cycles

Voltage dips andinterruptions

SafetyUL 61010-1:2012/R: 2018-11; CAN/CSA-C22.2 No. 61010-1:2012/A1:2018-11.
UL 61010-2-030:2018; CAN/CSA-C22.2 No. 61010-2-030:2018.

RoHS

EU 2015/863

SDS6 000L Data She et21

Power
Supply8-channel

4-channel

Input Voltage& Frequency

100 ~ 240 Vrms 50/60Hz
100 ~ 120 Vrms 400 Hz

Power
consumption

380 W max., 240 W typical, 8 W typicalin standby mode

193 W max., 120 W typical, 4 W typicalin standby mode

Mechanical8-channel

4-channel

Dimensions

Length × Height × Width =395 mm × 86 mm × 414 mm
Including rear I/O:
395 mm × 86 mm × 431 mm

Length × Height × Width =395 mm × 43 mm × 414 mm
Including rear I/O:
395 mm × 43 mm × 431 mm

Weight

Net Weight 9.1 kg

Net Weight 6.1 kg

22

SDS6000L Data SheetOrdering Information
Model

Description

SDS6208L

2 GHz bandwidth, 5 GSa/s sample rate, 8-bit, 500 Mpts memory depth, 8-channel

SDS6204L

2 GHz bandwidth, 5 GSa/s sample rate, 8-bit, 500 Mpts memory depth, 4-channel

SDS6108L

1 GHz bandwidth, 5 GSa/s sample rate, 8-bit, 500 Mpts memory depth, 8-channel

SDS6104L

1 GHz bandwidth, 5 GSa/s sample rate, 8-bit, 500 Mpts memory depth, 4-channel

SDS6058L

500 MHz bandwidth, 5 GSa/s sample rate, 8-bit, 500 Mpts memory depth, 8-channel

SDS6054L

500 MHz bandwidth, 5 GSa/s sample rate, 8-bit, 500 Mpts memory depth, 4-channel

Standard Accessories

Quantity

USB cable

1

Quick start

1

Passive probe

1/channel

Certificate of calibration

1

Wireless mouse

1

Power cord

1

Optional Accessories

Part No.

16-channel logic probe

SPL2016

Power Analysis deskew fixture

DF2001A

STB3 demo signal source

STB3

High-speed passive probe

SP6150A

High-speed active probe

SAP1000, SAP2500

High voltage probe

HPB4010High-speed differential probe

SAP2500D
DPB1300 / DPB4080 / DPB5150 / DPB5150A /DPB5700 / DPB5700A
CPL5100/CP4020/CP4050/CP4070/CP4070A
CP6030/CP6030A/CP6150/CP6500
SCP5030/SCP5030A/SCP5150/SCP5150A

High voltage differential probe
Current probe
Power rail probe

SAP4000P

64-channel synchronization distributor

SYN64

Options

Part No.

Waveform generator (software)

SDS6000L-FG

Power Analysis (software)

SDS6000L-PA

Eye Diagram/Jitter Analysis (software)

SDS6000L-EJ

I2S trigger & decode (software)
MIL-STD-1553B trigger & decode(software)
FlexRay trigger & decode (software)

SDS6000L-I2S

SDS6 000L Data She et

SDS6000L-1553B
SDS6000L-FlexRay

23

CAN FD trigger & decode (software)

SDS6000L-CANFDSENT trigger & decode (software)

SDS6000L-SENT

Manchester decode (software)

SDS6000L-Manch

24

SDS6000L Data SheetAbout SIGLENTSIGLENT is an international high-tech company, concentrating on R&D, sales,
production and services of electronic test & measurement instruments.

SIGLENT first began developing digital oscilloscopes independently in 2002.
After more than a decade of continuous development, SIGLENT has extended

its product line to include digital oscilloscopes, isolated handheld
oscilloscopes,

function/arbitrary

waveform

generators,

RF/MWsignalgenerators, spectrum analyzers, vector network analyzers, digital multimeters,
DC power supplies, electronic loads and other general purpose test
instrumentation. Since its first oscilloscope was launched in 2005, SIGLENT
has become the fastest growing manufacturer of digital oscilloscopes. We
firmly believe that today SIGLENT is the best value in electronic test &measurement.

Headquarters:
SIGLENT Technologies Co., Ltd
Add: Bldg No.4 & No.5, Antongda Industrial

Zone, 3rd Liuxian Road, Bao'an District,
Shenzhen, 518101, China
Tel: + 86 755 3688 7876
Fax: + 86 755 3359 1582
Email: sales@siglent.com
Website: int.siglent.com

North America:
SIGLENT Technologies America, Inc
6557 Cochran Rd Solon, Ohio 44139
Tel: 440-398-5800
Toll Free: 877-515-5551
Fax: 440-399-1211
Email: info@siglentna.com
Website: www.siglentna.com

Europe:SIGLENT Technologies Germany GmbH
Add: Staetzlinger Str. 70
86165 Augsburg, Germany
Tel: +49(0)-821-666 0 111 0
Fax: +49(0)-821-666 0 111 22
Email: info-eu@siglent.com
Website: www.siglenteu.com

Follow us on
Facebook: SiglentTech