SDS1000DL+/CML+

DataSheet-2020.02

Series Digital
Oscilloscope

SDS1000DL+/CML+ Series Digital Oscilloscope

Key Features

SDS1052DL+
SDS1072CML+
SDS1102CML+
SDS1152CML+

150 MHz, 100 MHz, 70 MHz, 50 MHz bandwidth models

Real-time sampling rate up to 1 GSa/s, Equivalent-time sampling
rate up to 50 GSa/s

Memory Depth up to 2 Mpts

Trigger types: Edge, Pulse, Video, Slope, Alternate
Waveform math functions:+, -, *, /, FFT
6 digital frequency counter

Product overview

Supports Multi-language display and embedded online help

SDS1000DL+/CML+ series is a dual-channel universal digital
oscilloscope, available in 50 MHz, 70 MHz, 100 MHz and 150 MHz
bandwidth models. It includes a 2 Mpts memory depth that helps
to ensure accurate waveform resolution and to capture longer
signal lengths. With its 7 inch TFT-LCD (800*480) screen, there isadequate screen space to help better see and analyze waveform
details. Along with a 1 GSa/s sampling rate, the SDS1000CML+
supports 32 parameters measurements and common
mathematical operations to speed up complex / repetitive
measurements.

Screensaver from 1 minute to 5 hours

Digital filter and waveform recorder function
Shortcut storage function key

7 inch TFT-LCD display with 800 * 480 resolution

Multiple interfaces: USB Host, USB Device (USBTMC), LAN (VXI11), Pass / FailLAN added

Models and Key Specifications
ModelSDS1052DL+

SDS1072CML+

Bandwidth

50 MHzChannels

2+EXT

Trigger Types

Edge, Pulse, Video, Slope, Alternate

Sampling Rate (Max.)

Memory Depth (Max.)

I/O

Probe (Std)Display

Net Weight

WWW.SIGLENT.COM

500 MSa/s

32 Kpts

70 MHz

1 GSa/s

2 pcs passive probe, PB470

2.5 Kg

SDS1152CML+

2 pcs passive probe, PP510

2 pcs passive probe, PP215

100 MHz

150 MHz2 MptsUSB Host, USB Device, LAN, Pass/Fail7 inch TFT LCD (800x480)

SDS1102CML+SDS1000DL+/CML+ Series Digital Oscilloscope

Function & Characteristic
Memory Depth up to 2 Mpts

Normal Memory (40 Kpts)

Long Memory (2 Mpts)

Using the long memory mode, users are able to use a higher sampling rate to capture more of the signal, and quickly zoom to focus on the area of interest.

32 parameters auto measurements and 5
parameters display

The SDS1000DL+/CML+ support voltage, time and delay measurement
types, with a total of 32 different parameters. The user is able toselect five measurements to display on the screen. All measurement
parameters can also be displayed simultaneously.

Pass/Fail Function

Zoom Function

Zoom can extend a partial segment of the waveform, giving the user

not only an overview of the whole signal but also a detailed view of thezoomed-in segment. The Zoom feature is a convenient way to locate a
specific segment of a signal while zooming in to see the details.

Math Function

With easy to generate user-defined test templates, the SDS1000DL+/

SDS1000DL+/CML+ provides 5 kinds of math operation: +, -, *, /, FFT,

trace making it suitable for long-term signal monitoring or automatedwindows or both signals appearing on the full screen.

CML+ compares the current measured trace to the template mask
production line testing.

supporting channel waveform and FFT waveform in either split display

WWW.SIGLENT.COM

SDS1000DL+/CML+ Series Digital Oscilloscope

Characteristics
Digital Recorder

The digital recorder is able to record data in real-time and without any

dead time. SDS1000DL+/CML+ supply 7 M of memory for the recorderReplaying the data for user to observe and analyze.

and support a USB disk.

Embedded Online Help

Abundant interfaces

Supports Multi-language display and embedded online help, familiarizes
the user with all the functions of in a short time.

Specifications
Acquire System

Real-time Sampling Rate
Memory Depth
Acquire Mode
Average

Waveform interpolation

Input

SDS1052DL+ ：32 Kpts
SDS1072CML+/SDS1102CML+/SDS1152CML+ : 40 Kpts (Normal Mode) ; 2 Mpts (Long Memory Mode)
Normal, Peak Detect, Average

Averages: 4, 16, 32, 64, 128, 256
Sinx,X
2

Impedance

DC: (1 MΩ±2%)||(18 pF ±3 pF)

Max. Input voltage
Channel Isolation
Probe attenuator

WWW.SIGLENT.COM11), Pass / Fail.

SDS1052DL+ ：500 MSa/s
SDS1072CML+/SDS1102CML+/SDS1152CML+ : 1 GSa/s

ChannelCoupling

SDS1000DL+/CML+ support USB Host, USB Device (USBTMC), LAN (VXI-

DC, AC, GND
400 V , 1 MΩ> 100:1

1 X, 10 X, 50 X, 100 X, 500 X , 1000 X

Horizontal System
Timebase Scale
Channel Skew

Display Format

Timebase Accuracy
Scan Mode

Vertical System
Bandwidth (-3 dB)
Vertical Resolution

Vertical Scale (Probe 1 X)
Offset Range (Probe 1 X)
Bandwidth Limit

Bandwidth Flatness
Low Frequency Response(AC-3 dB)
NoiseDC Gain Accuracy
DC Measurement Accuracy
Rise timeOvershoot (500 ps Pulse)

Trigger System
Trigger Mode

Trigger Level Range
Hold off Range

Trigger Coupling

Trigger Sensitivity
Trigger Displacement

Edge Trigger
Slope

SourceSlope Trigger

SDS1000DL+/CML+ Series Digital Oscilloscope
150 MHz 2.5 ns/div - 50 s/div
100 MHz 2.5 ns/div - 50 s/div
70 MHz 5.0 ns/div - 50 s/div
50 MHz 5.0 ns/div - 50 s/div<500 ps

Y-T, X-Y, Scan
±50 ppm

100 ms/div ~ 50 s/div
150 MHz (SDS1152 CML+)
100 MHz (SDS1102 CML+)
70 MHz (SDS1072 CML+)
50 MHz (SDS1052 DL+)
8 bit

2 mV/div - 10 V/div (1-2-5 )2 mV - 200 mV: ± 1.6 V; 206 mV ~ 10 V: ± 40 V
20 MHz ±40%

DC - 10%(BW): ± 1 dB
10% - 50%(BW): ± 2 dB
50% - 100%(BW): + 2 dB/-3 dB≤ 10 Hz (at input BNC)STDEV ≤ 0.6 div ( ≥ 5 mV/div)
STDEV ≤ 0.7 div (2 mV/div)≤ ±3.0%: 5 mV/div ~10 V/div≤ ±4.0%: ≤ 2 mV/div± [3%× (|reading|+|offset|) +1%×|offset| +0.2 div+2 mV] , ≤ 100 mV/div±[3%× (|reading|+|offset|) +1%×|offset| +0.2 div+100 mV] , >100 mV/div
Typical 2.3 ns (SDS1152 CML+)
Typical 3.5 ns (SDS1102 CML+)
Typical 5.0 ns (SDS1072 CML+)
Typical 7.0 ns (SDS1052 DL+)
<10%

Auto, Normal, Single

Internal: ±6 divisions from center of screen
EXT: ±1.2 V
EXT/5: ±6 V
100 ns ~ 1.5 s

AC, DC, LF Rej, HF Rej

1 Divisions: DC-10 MHz
1.5 Divisions: 10 MHz - Max BW

Pre-trigger: Memory depth/ (2*sampling)
Delay Trigger: 260 div
Rising, Falling, Rising & Falling
CH1/CH2/EXT/(EXT/5)/AC LineSlope

Rising, Falling

SourceCH1/CH2

Limit Range
Time Range

Pulse Trigger<, >, =20 ns ~ 10 s

Polarity+wid , -wid

SourceCH1/CH2

Limit Range
Pulse Range

Video Trigger<, >, =20 ns - 10 s

Signal Standard

NTSC, PAL/SecamTrigger condition

odd field, even field, all lines, line num

SourceCH1/CH2

WWW.SIGLENT.COM

SDS1000DL+/CML+ Series Digital Oscilloscope

Measure SystemSource

CH1, CH2

Vertical (Voltage)

Vmax

Highest value in input waveform

VppDifference between maximum and minimum data values

Measurement Parameters (32 Types)
VminVamp
Vtop

Vbase
Mean

VmeanVrms

Crms
FOVFPREROV
Horizontal (Time)

RPRE

PeriodFreq+Wid-Wid

Rise Time
Fall Time
Bwid+Dut-Dut

WWW.SIGLENT.COM

Value of most probable lower state in a bimodal waveform
Average of all data values

Average of data values in the first cycle (Condition: there is an entire period)
Root mean square of all data values

Root mean square of all data values in the first cycle (Condition: there is an entire period)
Overshoot after a falling edge; (base-min)/Amplitude

Overshoot before a falling edge; (max-top)/Amplitude
Overshoot after a rising edge;(max-top)/Amplitude

Overshoot before a rising edge; (base-min)/Amplitude

Period for every cycle in waveform at the 50% level ,and positive slope

Frequency for every cycle in waveform at the 50% level, and positive slope
Width measured at 50% level and positive slope

Width measured at 50% level and negative slope
Duration of rising edge from 10-90%

Duration of falling edge from 90-10%

Time from the first rising edge to the last falling edge, or the first falling edge to the last rising edge at the 50% crossing
Ratio of positive width to period

Ratio of negative width to period

FRF

Time from the first rising edge of channel A ,to the first falling edge of channel B

FFR
FFF

LRR
LFR

Counter

Value of most probable higher state in a bimodal waveform

Calculates the phase difference between two edges (Condition: there is an entire period)

LRFCursors

Difference between top and base in a bimodal signal ,or between max and min in an unimodal signal

Phase
FRR

Delay

Lowest value in input waveform

LFF

Time between the first rising edges of the two channels

Time from the first falling edge of channel A ,to the first rising edge of channel B

Time from the first falling edge of channel A ,to the first falling edge of channel B

Time from the first rising edge of channel A ,to the last rising edge of channel B (Condition: there is an entire period)

Time from the first rising edge of channel A, to the last falling edge of channel B (Condition: there is an entire period)
Time from the first falling edge of channel A, to the last rising edge of channel B (Condition: there is an entire period)
Time from the first falling edge of channel A, to the last falling edge of channel B

Manual mode, Track mode and Auto mode
Hardware Counter (Resolution 1 Hz)SDS1000DL+/CML+ Series Digital Oscilloscope

Math Function
Operation+ , - , * , / , FFTFFT display

Full Screen, Split

FFT

Save/Recall
TypeI/OStandard I/OPass/Fail

Display (Screen)

Rectangular, Blackman, Hanning, Hamming

Setting, Waveform, Bmp, CSV

2 refs, 20 settings, 10 waveforms internal
Save to USB disk

USB Host, USB Device, LAN, Pass/Fail3.3 V TTL Output

Display Type

7 inch TFT-LCD

Display Color

24 bitDisplay Resolution
Contrast (Typical)
Backlight

Wave display range

Wave Display Mode
PersistMenu Display
Screen-Saver
Color mode
Language

Environments
Temperature
Humidity
HeightPower Supply
Input

Power

Mechanical
Dimensions
Weight

800×480
500:1300 nit

8 x 16 div

Dots, Vectors

Off, 1 s, 2 s, 5 s, Infinite

2 sec, 5 sec, 10 sec, 20 sec, Infinite

Off, 1 min, 2 min, 5 min, 10 min, 15 min, 30 min, 1 hour, 2 hour, 5 hour
Normal , Invert

English, Simplified Chinese, Traditional Chinese, Arabic, French, German, Russian, Portuguese Spanish, Japanese, Korean,ItalianOperating: 10 ℃ ～ +40 ℃
Non-operating: -20 ℃ ～ +60 ℃

Operating: 85% RH, 40 ℃ , 24 Hours
Non-operating: 85% RH, 65 ℃ , 24 Hours
Operating: ≤ 3000 m
Non-operating: ≤ 15,266 m

100 ~ 240 Vrms 50/60 Hz
100 ~ 120 Vrms 400 Hz
50 W Max

Length 323.1 mm
Width 135.6 mm
Height 157 mm
N.W: 2.5 Kg

WWW.SIGLENT.COM

SDS1000DL+/CML+ Series Digital Oscilloscope

Probes & Accessories
TypePassive Probe

Model
PB470
PP510
PP215

CP4020PictureSpecifications
70 MHz Bandwidth

1 X/10 X,1 M/10 Mohm, 300 V/600 V
100 MHz Bandwidth

1 X/10X, 1 M/10 Mohm, 300 V/600 V
200 MHz Bandwidth

1 X/10X,1 M/10 Mohm, 300 V/600 V
Bandwidth: 100 KHz, Max. continuous current: 20 Arms, Peak current: 60 A
Switch Ratio: 50 mV/A, 5 mV/A, Accuracy: 50 mV/A (0.4A-10ApK)±2%,5 mV/A (1A-60ApK) ±2%, 9 V battery source

Bandwidth: 1 MHz; Maximum continuous current 50 Arms;CP4050

Peak current 140 A; Switching ratio: 500 mV/A; 50 mV/A;

DC measurement accuracy: 500 mV/A (20 mA-14 ApK) ±3%±20 mA;
50 mV/A (200 mA-100 ApK )±4%± 200 mA;

50 mV/A (100 A-140 ApK)±15% max; 9 V battery-powered
CP4070

Bandwidth: 150 KHz; Maximum continuous current 70 Arms;
Peak current 200 A;Switching ratio: 50 mV/A; 5 mV/A;

DC measurement accuracy: 50 mV/A (0.4 A-10 ApK) ±2%,±5 mV/A (1 A-200 ApK)±2%; 9 V battery-powered

Bandwidth: 300 KHz; Maximum continuous current 70 Arms;
CP4070A

CP5030CP5030A

CP5150CP5500

DPB4080

DPB5150

DPB5150A

WWW.SIGLENT.COM

DC measurement accuracy: 100 mV/A (50 mA-10 ApK) ±3%±50 mA;
10 mV/A (500 mA-40 ApK) ±4%±50 mA;

10 mV/A (40 A-200 ApK) ±15%max; 9 V battery-powered

Current Probe

High Voltage
Differential
Probe

Peak current 200 A;Switching ratio: 100 mV/A;10 mV/A;

Bandwidth: 50 MHz; Maximum continuous current 30 Arms;
Peak current 50 A;Switching ratio: 100 mV/A, 1 V/A;
AC/DC measurement accuracy: 1 A (±1%±1 mA);

100 mV/A (±1%±10 mA); Standard DC12 V/1.2 A power adapter
Bandwidth: 100 MHz; Maximum continuous current 30 Arms;
Peak current 50 A;Switching ratio: 100 mV/A, 1 V/A;
AC/DC measurement accuracy: 1 A (±1%±1 mA);

100 mV/A (±1%±10 mA); Standard DC12 V/1.2 A power adapter
Bandwidth: 12 MHz; Maximum continuous current 150 Arms;
Peak current 300 A;Switching ratio: 100 mV/A, 1 V/A;

AC/DC measurement accuracy: 100 mV/A(±1%±1 mA);

10 mV/A (±1%±10 mA); Standard DC12 V/1.2 A power adapter
Bandwidth: 5 MHz; Maximum continuous current 500 Arms; Peak current750
A; Switching ratio: 100 mV/A, 10 mV/A;

AC/DC measurement accuracy: 100 mV/A (±1%±1 mA);

10 mV/A (±1%±10 mA); Standard DC12 V/1.2 A power adapter
Bandwidth: 50 MHz;

Maximum input differential voltage 800 V (DC + Peak AC);

Range selection (attenuation ratio):10 X/100 X; Accuracy: ±1%;
Standard DC 9 V/1 A power adapter
Bandwidth: 70 MHz;

Maximum input differential voltage 1500 V (DC + Peak AC);

Range selection (attenuation ratio): 50 X/500 X; Accuracy: ±2%;Standard 5 V/1 A USB power adapter
Bandwidth: 100 MHz;

Maximum input differential voltage 1500 V (DC + Peak AC);

Range selection (attenuation ratio): 50 X/500 X; Accuracy: ±2%;Standard 5 V/1 A USB power adapter

SDS1000DL+/CML+ Series Digital Oscilloscope

Type

High Voltage
Differential
Probe

ModelPicture

DPB5700

DPB5700A

High Voltage
Probe

HPB4010Isolated front
end

ISFE

Demo board

STB3

Ordering information

Specifications
Bandwidth: 70 MHz;

Maximum input differential voltage 7000 V (DC + Peak AC);
Range selection (attenuation ratio): 100 X/1000 X;

Accuracy: ±2%; Standard 5 V/1 A USB power adapter
Bandwidth: 100 MHz;

Maximum input differential voltage 7000 V (DC + Peak AC);
Range selection (attenuation ratio): 100 X/1000 X;

Accuracy: ±2%; Standard 5 V/1 A USB power adapter
Bandwidth: 40 MHz; Maximum measurement voltage DC: 10 KV;
AC (rms) : 7 KV (sine) ;AC (Vpp) :20 KV (Pulse);
attenuation ratio1:1000; Accuracy: ≤ 3%
USB 5 V power supply, plug and play,

the maximum input voltage 600 Vp-p, floating test. Work with oscilloscopes.

Optional accessories For experimental teaching and product demos

Description

Model

70 MHz, 2 CH, 1 GSa/s (Max.) , 2 Mpts, 7 inch (800*480) LCD

SDS1072CML+

50 MHz, 2 CH, 500 MSa/s (Max.) , 32 Kpts, 7 inch (800*480) LCD
100 MHz, 2 CH, 1 GSa/s (Max.) , 2 Mpts, 7 inch (800*480) LCD
150 MHz, 2 CH, 1 GSa/s (Max.) , 2 Mpts, 7 inch (800*480) LCD

Standard Accessories

SDS1052DL+

SDS1102CML+
SDS1152CML+

USB Cable -1

Quick Start -1

Certificate of Calibration -1
Passive Probe -2
Power Cord -1

Optional Accessories
Isolated Front End

ISFEHigh Voltage Probe

HPB4010

STB Demo board
Current Probe

Differential Probe

STB3

CP4020/CP4050/CP4070/CP4070A/CP5030/CP5030A/
CP5150/CP5500
DPB4080/DPB5150/DPB5150A/DPB5700/DPB5700A

WWW.SIGLENT.COM

SDS1000DL+/CML+Series Digital
Oscilloscope

About SIGLENTSIGLENT is an international high-tech company, concentrating on R&D, sales,production and services of electronic test & measurement instruments.

SIGLENT ﬁrst began developing digital oscilloscopes independently in 2002.
After more than a decade of continuous development, SIGLENT has extended
its product line to include digital oscilloscopes, function/arbitrary waveformgenerators, RF generators, digital multimeters, DC power supplies, spectrum
analyzers, vector network analyzers, isolated handheld oscilloscopes,
electronic load and other general purpose test instrumentation. Since its ﬁrst
oscilloscope, the ADS7000 series, was launched in 2005, SIGLENT has become
the fastest growing manufacturer of digital oscilloscopes. We ﬁrmly believethat today SIGLENT is the best value in electronic test & measurement.

Headquarters:
SIGLENT Technologies Co., Ltd
Add: Bldg No.4 & No.5, Antongda IndustrialZone, 3rd Liuxian Road, Bao'an District,Shenzhen, 518101, China
Tel: + 86 755 3688 7876
Fax: + 86 755 3359 1582
Email: sales@siglent.com
Website: int.siglent.com
USA:
SIGLENT Technologies America, Inc
6557 Cochran Rd Solon, Ohio 44139
Tel: 440-398-5800
Toll Free: 877-515-5551
Fax: 440-399-1211
Email: info@siglent.com
Website: www.siglentna.com

Europe:SIGLENT Technologies Germany GmbH
Add: Liebigstrasse 2-20, Gebaeude 14,22113 Hamburg Germany
Tel: +49(0)-819-95946
Fax: +49(0)-819-95947
Email: info-eu@siglent.com
Website: www.siglenteu.com

Follow us on
Facebook: SiglentTech