SPEC SHEETSGD

Current Loop Supply (1-output)
Features:

SGD (1-input 1-output ) converts DC current input into isolated DC
voltage/current. This unit can be used as the power source for a 2-wiretransmitter (sensors, etc.). Indication displays are equipped.

Functions: Square root extraction function ‘SMART’ transmitter compatible(Shunt resistor 50Ω + Check resistance 250Ω）
 Sensor correction (Input value correction)
 Normal/Reverse mode
 Filter time constant Manual mode Output type can be changed.

Model: SGD – A0 ② - ③ - ④Model name
Input 1
Output 1
Power supply
OptionINPUT 1
DC current

A0

4 to 20mA(Built-in 50Ω shunt resistor)③ Power supply
0
1

100 to 240V AC 50/60Hz
24V AC/DC 50/60Hz
For SGDL (with serial communication function),
only [0: 100 to 240V AC] is selectable.

④ Option

DC Voltage

DC Current② OUTPUT 1
12345A
BCDEFG

4 to 20mA (allowable load resistance 750Ω max.)
0 to 20mA (allowable load resistance 750Ω max.)※1
0 to 16mA (allowable load resistance 900Ω max.)※1
2 to 10mA (allowable load resistance 1500Ω max.)
0 to 10mA (allowable load resistance 1500Ω max.)※1
0 to 10mV (allowable load resistance 10kΩ min.)※20 to 100mV(allowable load resistance 100kΩ min.)※2
0 to 1V (allowable load resistance 1000Ω min.)※2
0 to 5V (allowable load resistance 5000Ω min.)※2
1 to 5V (allowable load resistance 5000Ω min.)0 to 10V (allowable load resistance 10kΩ min.)※2-5 to 5V (allowable load resistance 10kΩ min.)

0
1
2
3

No option needed
Multi-rotation trimmer
Moisture-proof treatment
Multi-rotation trimmer + Moisture-proof treatment※1: 0mA or less: Out of base accuracy.
※2: 0V or less: Out of base accuracy.

SHINKO TECHNOS CO., LTD. Tel: +81-72-727-6100
URL: http://www.shinko-technos.co.jp/e/E-mail: overseas@shinko-technos.co.jp

SPEC SHEET
Performance

Base accuracy (at 25℃)
Display accuracy
Temperature coefficient
Response time
Insulation resistanceDielectric strength±0.1% of each input span
Base accuracy ±1 digit± 0.015%/℃ (0 to 10mV output:±0.02%/℃)
0.5 sec. max. (0→90%)
100MΩ minimum, at 500V DC
2.0kV AC for 1 minute

General specifications
Input
Power supply for 2-wire transmitter
Zero adjustment range
Power supply
Power consumption
Operating temperature, humidity
Storage temperature
Material
Mounting method
External dimensions
WeightInput resistance: Approx. 300Ω
(Shunt resistor 50Ω + Check resistance 250Ω）
Output voltage: 24 to 28V DC (When load current is 20mA)
Ripple voltage: Within 200mV (When load current is 20mA)
Max. load current: 25mA
-5 to 5% (Adjustable from front panel.)
Span adjustment range
95 to 105% (Adjustable from front.)
100 to 240V AC 50/60Hz
85 to 264V AC
Allowable voltage range
24V AC/DC 50/60Hz
20 to 28V AC/DC
100 to 240 V AC: Approx. 10 VA max. (SGDL: Approx. 11 VA max.)
24 V AC: Approx. 7 VA max.
24 V DC: Approx. 4 W max.
-10 to 55℃ (Non-condensing, No icing), 35 to 85%RH (Non-condensing)-10 to 60℃
Case: Flame-resistant resin, Black color, Panel: Polycarbonate
DIN rail22.5(W) x 89(H) x 70(D)mm (without socket)
Approx. 78g (without socket)

Measurement range

Terminal arrangement

Input code

Measurement range

ResolutionA0-1999 to 9999 ※1※ Decimal point position change and scaling are possible.

PWR ⑬ - ⑭
OUT ⑨ - ⑫A ①-④
24V ④ - ⑧RS-485

Circuit configuration

Power supply voltage
100 to 240V AC, 24V AC/DCOutput
DC current input
Power supply for 2-wire transmitter
Serial communication RS-485(For SGDL)(8P Socket)

When using as a current loop supply

When using as an isolator or square root extractor

SHINKO TECHNOS CO., LTD. Tel: +81-72-727-6100
URL: http://www.shinko-technos.co.jp/e/E-mail: overseas@shinko-technos.co.jp

20190507