1 OPTICAL MEASURING INSTRUMENTS....................................................... p.2
2 BIT ERROR RATE TESTERS (BERT)/OSCILLOSCOPES................................. p.4
3 TRANSPORT AND ETHERNET TESTING..................................................... p.5
4 MOBILE/WIRELESS COMMUNICATION MEASURING INSTRUMENTS.......... p.6
5 SIGNAL ANALYZERS/SPECTRUM ANALYZERS.......................................... p.11
6 VECTOR NETWORK ANALYZERS.............................................................. p.14
7 SIGNAL GENERATORS............................................................................. p.17
8 RF MICROWAVE MEASURING INSTRUMENTS.......................................... p.18

Network Master Pro
Mainframe MT 1000 A
OTDR Module MU100020 A/MU100021 A/MU 100022 A

1310 nm/ 1550 nm SMF, 1310 /1550 / 850 / 1300 nm SMF/MMF, 1310 / 1550 / 1625 nm SMF

Mobile Network I&M● All-in-one OTDR, light source, optical power meter (standard), visible light source (option)● CPRI/OBSAI measurement with simultaneously installed 10G (MU100010A)/100G (MU100011A)
Multirate Module● Optical connector inspection with IEC 61300-3-35 pass/fail● Graphical summary and pass/fail evaluation using Fiber Visualizer function● Intuitive touch-screen operation

OPTICAL MEASURING INSTRUMENTS
2The OTDR module lineup includes the MU100021A for OTDR measurements of both SM and MM
fibers in high demand by the mobile network I&M, plus the MU100020A/MU100022A for OTDR
measurements of SM fiber used by PON networks and long-range measurements in Core/Metro networks.

ACCESS Master™
MT 9085 series

850 nm/ 1300 nm (MM), 1310 /1490 / 1550 / 1625 / 1650 nm (SM)All-in-One Solution for Optical Fiber Construction and Maintenance of Core,
Metro and Access Networks● 8-inch LCD with easy visibility even in direct sunlight● Better work efficiency with synergy of LCD touchscreen, rotary knob, and dedicated hard keys
● Easy-to-Use Fiber Visualizer function for simple fiber path analysis
The MT9085 series is a compact handheld all-in-one tester for performing optical pulse tests,
optical loss/power measurements, and optical fiber end-face inspections. It has a wide variety
of applications, ranging from installation and maintenance (I&M) of trunk fibers (Core, Metro,
Mobile Fronthaul, Mobile Backhaul) to troubleshooting Access networks, such as breaks in drop cables.

Network Master™
Mainframe MT 9090 A
μOTDR Module™ MU909014 /MU909015

1310 / 1490 / 1550 nm plus filtered 1650 nm or 1625 nm

Field Optical Testing Redefined● High-performance OTDR in a pocket-size package with unique battery operation● Tri-wavelength OTDR for both installation and maintenance1310/1490/1550 nm plus filtered 1650 nm or 1625 nm● Built-in PON power meter, loss test set and light source function● "Fiber Visualizer" mode simplifies operation, no OTDR knowledge needed● Bluetooth, WLAN and Ethernet connectivity

Optical Loss Tester/Light Source/Optical Power Meter
CMA 5 series

850 nm/ 1300 nm (MM), 1310 /1490 / 1550 /1625 nm (SM)For Optical Fiber Installation and Maintenance● Built-in light source and power meter (Optical Loss Tester)● Two wavelengths at one port (Light Source)● Level measurement up to +23 dBm (Optical Power Meter)
The compact and durable design of the CMA5 series make these instruments the ideal
combination of light source and optical power meter for measuring optical power when
installing and servicing optical fiber cables.

Video Inspection Probe
Autofocus Video Inspection Probe G 0382 A
Video Inspection Probe G 0306 B
Optical Connector End Face Inspection● Fully automated one-button operation (G0382A)● Supported pass/fail analysis with the IEC61300-3-35 standard● Wide range of adaptors available
The Video Inspection Probe (VIP) application for Anritsu field testing
platforms gives operators a safe, easy way to analyze and document
connector conditions.

G 0306 B

OPTICAL MEASURING INSTRUMENTS

The MU909014/15 series for the MT9090A from Anritsu finally addresses this need by providing all of the features and performancerequired for installation and maintenance of optical fibers in a compact.
The MT9090A represents an unmatched level of value and ease of use, while not compromising performance. Data sampling of fivecentimeters, dead zones of less than 0.8-meter and dynamic range up to 38 dB ensure accurate and complete fiber evaluation of any
network type – premise to access, metro to core…including PON-based FTTx networks featuring up to a 1 × 64 split.

G0382 A

Optical Spectrum Analyzer
MS9740 B

600 nm to 1750 nm

Reduces Measurement Time and Improves Production Efficiency● Wavelength sweeping time <0.35 s● Dedicated applications for evaluating active optical devices● Excellent cost performance● Dynamic range performance ≤58 dB (0.4 nm from peak wavelength)● 30 pm minimum resolution
The MS9740B reduce the measurement processing times by up to half compared tothe earlier model while assuring high performance and complete test menus brings
higher-efficiency inspection of active optical devices.

Coherent OTDR
MW 90010 A
Measures Submarine Cables up to 12 ,000 km Long● Fault detection with 10 m distance resolution● Compact and lightweight all-in-one design for on-site portability
320 (W) × 177 (H) × 451 (D) mm, <17 kg● Simple and easy touch-panel operation for easy first-time use by any operator●W ide dynamic range supporting fault detection and troubleshooting of submarine cables
with repeaters at 80 km or longer intervals
The MW90010A is a measuring instrument for detecting faults in ultra-long opticalsubmarine cables of up to 12,000 km including multiple repeaters (EDFAs).

3

Signal Quality Analyzer-R
MP1900 A
Support 400 GbE and PCIe Gen 4 / 5 .
All-in-One Support for Evaluating Next-Generation NRZ/PAM4 Network Interfaces and High-Speed Serial Buses

BIT ERROR RATE TESTERS (BERT)/OSCILLOSCOPES
4● All-in-one support for both high-speed Ethernet and PCI Express interface tests● Easily configured and easy-to-use all-in-one 64-GBaud PAM4 BER measurement
system requiring no external equipment● Receiver tests are supported by the built-in Protocol Awareness PCIe Link Training andLTSSM analysis functions● High-reproducibility measurements due to high waveform quality and high input
sensitivity● Supports true signal integrity analysis, such as bit error rate measurements, Jitter
Tolerance tests, etc.
The MP1900A is a high-performance BERT with excellent expandability for supporting
Physical layer evaluations of these high-speed interfaces. The all-in-one design is ideal
for early stage R&D evaluations of all interfaces covering ext-generation Ethernet
networks to bus interconnects.

BERTWave™
MP2110 A
For 100 G/200 G/ 400 G Multi-channel Optical Module/Device R&D and Manufacturing● BERT and sampling oscilloscope for up to 4ch installed one unit
● Supports analysis for both NRZ and PAM4 signals with high-spend, low-noise sampling
oscilloscope, and built-in CRU
With a built-in BERT (for Bit Error Rate measurements) and a sampling oscilloscope (for Eyepattern analysis) the All-in-one MP2110A is optimized for manufacturing 100G/200G/400G
optical modules. The MP2110A will improve optical module production efficiency andreduce manufacturing costs.

BERTWave™
MP2100 B
For 10 G/ 40 G Multi-channel Optical Module/Device R&D and Manufacturing● All-in-one BER and Eye-pattern analysis● Built-in 1ch to 4ch 12.5 Gbit/s BERT
● High-speed mask tests● Jitter 1 ps high-quality PPG and 10 mVp-p high-sensitivity ED
The all-in-one MP2100B has a built-in BER tester and sampling oscilloscope for runningsimultaneous BER tests and eye pattern analyses required for developing and manufacturing
modules. The number of BERT channels can be expanded to four, all supporting simultaneous
BER measurements. Additionally, the high sampling speed reduces the eye pattern measurement
time. Multi-channel optical modules, such as QSFP+, can be measured more efficiently using theMP2100B.

Network Master Pro
Mainframe MT 1000 A
10 G Multirate Module MU100010 A
100 G Multirate Module MU100011 A
All-in-One Transport Tester for Metro and Backhaul Network Installation and Maintenance● Supports testing from 1.5 Mbps to 100 Gbps
● Remote operation● Remote control (scripting)● Compact, lightweight design for maximum field portability

Network Master Pro
Mainframe MT 1000 A
CPRI RF Module MU100040 B
The MU100040 B CPRI RF module for the MT 1000 A provides added versatility to our new Front Haul testing platform● Displays LTE spectrum of ALU/Nokia, Ericsson and Huawei CPRI radios● Fast update rate to capture intermittent interferes● Spectrum pan and zoom for detailed analysis of interferes/Spectrogram display captures andholds data for intermittent interferes● 2 SFP slots for simultaneous uplink and downlink testing● Modular design for use with MT1000A OTDR and 10G/100G transport test modules
The MU100040B for the MT1000A adds CPRI RF measurements to Anritsu’s transport and fiber
test platform. The modular design of the MT1000A means that it can be configured just for CPRI
measurements or combined with the 10G/100G transport module and OTDR module to create
the most comprehensive and versatile fiber and transport tester available.

Network Master Flex
MT 1100 A

1 . 5 Mbps to 100 Gbps

TRANSPORT AND ETHERNET TESTING

The modular design of the Network Master Pro MT1000A platform makes it easy to support I&M
for different network configurations. Combining it with the MU100010A offers the necessaryfunctions for I&M of networks at speeds from 1.5 Mbps to 10 Gbps. Combining with the
MU100011A, it supports more interface standards than any other handheld transport tester on
the market such as CFP4/QSFP28, QSFP+, SFP28 (25GbE), SFP+/SFP and RJ45.

All-in-One Transport Tester for 100 G Core/Metro Network R&D and Manufacturing● Supports testing from 1.5 Mbps to 100 Gbps
● Up to 4 ports at all rates● Remote operation● Remote control (scripting)● Modular platform ensuring maximum return on investment
The all-in-one MT1100A supports all the latest communications network technologies.
Selecting and installing up to two modules from a range of three module options supports
all-in-one R&D and manufacturing tests of network and transport equipment operating at bit
rates from 1.5 Mbps to 100 Gbps.

Network Master™
Mainframe MT 9090 A
Gigabit Ethernet Module MU909060 A 1 /A 2 /A 3
Handheld Gigabit Ethernet Tester● Lightweight and compact unit (approx. 800 g)● Testing time reduced by the "Test Automator" creating a series of tests with pass/fail● Automated ITU-T Y.1564 and RFC 2544 testing including bidirectional path analysis service● Disruption time measurement ideal for testing VoIP and IPTV applications top talkers,
network attacks and finding the route course of an issue by "Channel Stats"
The portable and easy-to-use MU909060A offers versatile measurement functions supports
deployments and maintenances of Carrier Class Ethernet and LTE mobile backhaul networks.

5

Radio Communication Test Station
Mainframe MT 8000 A
RF TRx Measurement Software MX800010 A
Protocol Test Software MX800030 A
Rapid Test Designer Platform (RTD) MX800050 A
All-in-One 5 G Signaling and RF Tests

MOBILE/WIRELESS COMMUNICATION MEASURING INSTRUMENTS
6● Supports RF measurement and protocol testing from sub 6 GHz band to millimeter wave band
including Band n41● Combined with RF Chamber, Beam Management test with millimeter wave is supported● The modular architecture makes it possible to flexibly build a test environment suitable forvarious purposes● Interlock test that simulates 5G and LTE NSA (non-standalone) using existing LTE test environment is supported● Supports both SA (Standalone) and NSA (non-Standalone) on MT8000A. Possible to use existing LTE test environment for NSA
The MT8000A is a platform supporting both 5G RF measurement/protocol testing. This instrument strongly support the development ofchipset, UE etc. corresponding to the frequency bands of 5G communication used all over the world.

mmW OTA test environment for 5G NR
Shield Box MA 8161 A
RF Chamber MA 8171 A
CATR Anechoic Chamber MA 8172 A
For 5 G NR mmW testing● MA8161A: A
 mmW OTA connection environment can bebuilt on the desktop● MA8171A: R
 F/Protocol mmW OTA environment can bebuilt with this single unit
● MA8172A: 3
 GPP compliant CATR chamber for
testing 5G NR mmW devices

MA8161 A
You can choose from 3 mmW OTA test environments
according to your application, such as RF TRx test and Protocol test.

MA8171 A

MA8172 A

Signalling Tester/Rapid Test Designer (RTD)
MD8430 A/MX800050 A/MX786201 A
Powerful Support for Developing LTE/LTE-Advanced/LTE-Advanced Pro/ 5 G NSA Anchor Chipsets and Mobile UEs
GUI for Creating● Supports LTE-Advanced FDD/TDD Carrier Aggregation (CA) 2CCs, 3CCs, 4CCs, 5CCs and 6CCs
● One MD8430A supports CA handover, 4 × 4 MIMO and 8 × 4 MIMO
● Supports full-digital fading test
● Supports DL 1 Gbps (PHY: 2.0 Gbps), UL 300 Mbps throughput test
● Supports 5G NSA protocol testing is possible by using in combination with Radio Communication
Test Station MT8000A● Full development and analysis toolset cuts L1, L2 and L3 scenario development time and costs● Supports UMTS Release 10, HSPA Evolution, GSM/GPRS/EGPRS● Easy to create, and execute test cases. Easy network parameter management
The MD8430A is a key LTE-Advanced Pro base station simulator for developing LTE-Advanced Procompliant chipsets and mobile devices. Using its extensive experience in 3G markets, Anritsu has
developed the MD8430A as a powerful LTE-Advanced Pro R&D test solution to help developers bring
LTE/LTE-Advanced/LTE-Advanced Pro UEs to market as fast as possible.
The MX800050A/MX786201A is a GUI tool for creating and exacting test cases based on 3GPP,
including 5G NR, LTE/LTE-Advanced/LTE-Advanced Pro, W-CDMA/HSPA, GSM handover tests, that runs on the MT8000A/MD8430A.

Signalling Tester
MD8475 B

350 MHz to 6 GHz, 8 TX/ 4 RX

For General Wireless Device Application Tests● Scenario-less test environment● Supports most mobile communications technologies from LTE-Advanced to legacy (2G/3G)● Built-in IMS server enables to test VoLTE● Supports LTE-Advanced throughput test using build-in IP Traffic Generator (MD8475B only)● Function test of the smartphone such as RCS or WLAN offload● Supports automotive emergency call system (eCall, ERA-GLONASS), enables to verify thecommunication quality of in-vehicle module and device● Automated continuous 24/7 testing using SmartStudio Manager

Simple Conformance Test System
ME 7800 L
Simple configuration conformance test system supporting RF/RRM and Protocol testing by 1 platform● Support LTE RF/RRM and Protocol conformance testing● Support IoT testing (Cat-M, NB-IoT)● GCF and PTCRB validated conformance test system● Measurement functions for supporting R&D● Superior measurement reproducibility and stability● Support for Regional Frequency BandsThe ME7800L is the ideal system for introducing RF and Protocol Conformance
tests of 3GPP-compliant LTE mobile terminals and IoT devices.

5G NR Mobile Device Test Platform
ME 7834 NR
GCF/PTCRB, and Carrier Approved Test System for 5 G/4 G Mobile Protocol Testing● All-in-One 5G/4G NR Support for Protocol Conformance Tests and Carrier Acceptance Test
● Initial setup and training support● High test efficiency with reliable automation and easy GUI● Supports 3GPP defined bands from Sub-6 GHz to mm-Wave● Upgrade your current ME7834 system for 5G● Reduction of test time & cost by automatic continuous test execution function
The ME7834NR is an all-in-one test system to support 3GPP- based Protocol Conformance
Tests (PCT) and Carrier Acceptance Tests (CAT) for Sub 6 GHz and mmWave.
In addition to the 5G New Radio (NR) Technology in both Standalone (SA) and NonStandalone (NSA) modes, ME7834NR support LTE, LTE-Advanced (LTE-A), LTE-A Pro,
W-CDMA and GSM to strongly assist certification test of global 5G & 4G communication
device and chipset.

LTE-Advanced Mobile Device Test Platform
ME 7834 LA
GCF/PTCRB, and Carrier Approved Test System for LTE-Advanced Mobile Protocol Testing● No. 1 GCF and PTCRB approved test cases● Initial setup and training support● High test efficiency with reliable automation and easy GUI● Carrier approved test platform● Support LTE-Advanced by Single Rack

MOBILE/WIRELESS COMMUNICATION MEASURING INSTRUMENTS

The all-in-one MD8475B supports the full range of smartphone and wireless device tests; when used in combination with SmartStudio, it
handles all the complex functions and application tests required by manufacturers and vendors.

The ME7834LA supports quick and easy 3GPP TS 36.523, TS 34.229, TS 34.123 and TS 51.010 protocol
conformance tests of 3G/4G mobile systems as well as carrier acceptance testing for AT&T, Verizon,
T-Mobile USA, docomo, KDDI, SoftBank, TD-LTE NS-IOT and SK telecom.

7New Radio RF Conformance Test System
ME 7873 NR
5 G NR RF/RRM conformance system supporting both sub-6 GHz and mmWave band
● Industry-first test case validation● Test system realizing superior measurement reproducibility and stability● Measurement functions for efficient R&D● Multiple hardware configurations tailored to measurement requirements■ TRX Full/Performance/RRM● Tunable filtering supports multiple bands with no hardware upgrade● Upgradable from ME7873LA supporting W-CDMA, LTE and LTE-Advanced● Support both 5G NR Non-Standalone mode and Sanedalone mode

MOBILE/WIRELESS COMMUNICATION MEASURING INSTRUMENTS

ME7873NR is the RF/RRM conformance system complying with 3GPP TS 38.521
and TS 38.533 for both sub-6 GHz and mmWave band. ME7873NR won WorldFirst GCF Approval for 5G RF, and the number of Approved TCs is the top classin the market.
And ME7873NR supports Regulatory testing and RF testing for US Operator.

LTE-Advanced RF Conformance Test System
ME 7873 LA
RF/RRM Conformance Test System Supporting W-CDMA, LTE/LTE-A/LTE-A Pro by 1 platform● Industry-first test case validation● Test system realizing superior measurement reproducibility and stability● Measurement functions for efficient R&D● Multiple hardware configurations tailored to measurement requirements■ TRX/Performance/RRM● Tunable filtering supports multiple bands with no hardware upgrade● Inter-RAT handover capability■ LTE to GSM/UMTS/CDMA2000/TD-SCDMA● Support LTE-Advanced Pro, such as LAA and Cat-M/NB-IoT, Joint CA up to 4CA, 4 × 4 MIMO, andHPUE testing
ME7873LA is the world's first PTCRB-validated RF test platform for 5DL CA (June 2018).
And ME7873LA supports many test cases for LAA and 4Rx/4 × 4 MIMO.
The number of Approved TCs is top class in the market, and ME7873LA contributes to theevolution and development of mobile broadband service.

Radio Communication Analyzer
MT 8821 C

30 MHz to 3 .8 GHz/ 6 .0 GHz

All-in-One Integrated Wireless Tester for RF TRX measurement of LTE/LTE-Advanced
and Cat-M/NB-IoT, 5 G NSA Anchor operation● Supports multiple communications technologies from LTE-Advanced, IoT to legacy (2G/3G)● Easy-to-use GUI for 3GPP RF TRX tests. In addition, operation is made easy just by choosing
item numbers using the automatic test tool● Supports VoLTE tests with built-in IMS server● Upgradeable from MT8820C. Compatible functions, performance and remote commands
The MT8821C is a measuring instrument for mobile terminal developers; it is the successor to Anritsu’s popular MT8820C used by mobile
terminal and chipset vendors worldwide. As well as inheriting MT8820C technologies and know-how, the MT8821C adds support for new
functions, including DL CA 8 CC, LTE-U/LAA, DL 256QAM, DL 4 × 4 MIMO, UL CA, and LTE Cat-M/NB-IoT. Also, MT8821C supports 5G NSA
Anchor operation and call connection with UE by combining with MT8000A. As well as supporting RF measurements, the MT8821C also
supports other R&D tests ranging from RF calibration, inspection, and functional tests. Additionally, it supports a full range of evaluation
items for developing smartphones and communications module hardware with OTA and SAR test solutions customized by each vendor.

Vector Signal Generator
MG3710 E

100 kHz to 2 . 7 GHz/ 4 GHz/6 GHz

Supports the Evaluation of Wireless Communications Evolving into the 5 G● RF modulation bandwidth 160 MHz*/120 MHz● Pre-installed key waveform patterns● Waveform addition function
Adds and outputs two signals, such as wanted signal + interference signal orwanted signal + AWGN● One unit supports two RF outputs max., Ideal for multi-system evaluations● Supports BER test function*: When using MX370111A/MX370111A-002

8

The MG3710E supports various digital modulation signals, such as 5G, LTE/LTE-Advanced, GSM/GPRS/EDGE, W-CDMA/HSPA/HSPA
Evolution, TD-SCDMA, Digital Broadcast (ISDB-T, DVB-T/H, CMMB), GPS, Bluetooth, and WLAN for major communication systems.
It is ideal for tests of base stations, mobile terminals and devices.

Signal Analyzer
MS2690 A/MS2691 A/MS2692 A
MS2850 A
MS2830 A

50 Hz to 6 GHz/13 .5 GHz/ 26 . 5 GHz
9 kHz to 32 GHz/ 44 . 5 GHz
9 kHz to 3 .6 GHz/6 GHz/ 13 . 5 GHz

Supports All Key Communications Systems

The MS2690A/MS2691A/MS2692A/MS2850A/MS2830A have all the versatile built-in measurement functions needed for evaluating Tx
characteristics. Parameter setting is easy using pre-installed templates for each measurement standard. Installing measurement options
displaying numerical and graphical results supports modulation analysis of key communications systems. Various modulation signals
are output by the built-in waveform patterns of the Vector Signal Generator option for the MS2690A/MS2691A/MS2692A/MS2830A. The
IQproducer software makes it easy to generate and edit standards-based waveforms, such as 5G.

Wireless Connectivity Test Set
MT 8862 A

2 . 4 GHz to 2 . 5 GHz, 5 GHz to 6 GHz

Ideal for RF TRx Tests of WLAN Devices● Network Mode/Direct Mode Support
Network mode is not required to control chipset and use test
firmware for measurement under realistic operation conditions● Wide Connectivity Support
Connections are supported in the IEEE802.11a/b/g/n/ac/ax AP and
STA modes. Additionally, supporting securities● Measureable at specified Data Rate of DUT
Equipped with a control function to converge to the data rate you
specify the DUT, also it supports 11ac/ax● MIMO Measurement Support
2 × 2 MIMO Measurement function Installed on 11n/ac● Built-in IP Data Ports
Ethernet ports for IP data are built-in and IP continuity tests on 11 a/b/g/n/ac
Anritsu’s Wireless Connectivity Test Set is ideal for measuring the RF TRx characteristics at design and manufacturing inspection of WLAN
devices. It has a built-in Network Mode for measuring the performance of the WLAN DUT under realistic operation conditions to play a
key role in quality evaluation and improvement.

Universal Wireless Test Set
MT 8870 A

10 MHz to 3 .8 GHz/ 6 GHz

Designed to Maximize Production Throughput of Smartphones and Wireless Modules● All-in-one platform accommodating 4 test modules● Simultaneous measurement of 4 devices and parallel measurement of multiple wirelesscommunications systems in one device● 160 MHz measurement bandwidth as standard● Support multiple wireless standards
5G NR sub-6 GHz, LTE/LTE-Advanced, W-CDMA/HSPA, TD-SCDMA, GSM/EDGE,
CDAM2000/1xEV-DO, WLAN 802.11a/b/g/n/p (V2X)/ac/ax, Bluetooth v5.0, ZigBee, Z-Wave,
FM/RDS, GPS/Galileo/GLONASS/BeiDou, DVB-H, ISDB-T/ISDB-Tmm● Built-in audio analyzer and audio generator
The MT8870A is a test instrument from Anritsu that has been specifically designed for high volume manufacturing test of cellular andconnectivity wireless system. The MT8870A instrument mainframe can contain up to four TRX Test modules MU887000A/01A. Each module
has an integrated Vector Signal Generator (VSG) and Vector Signal Analyzer (VSA) to perform DUT transmitter and receiver RF tests.

MOBILE/WIRELESS COMMUNICATION MEASURING INSTRUMENTS● Analysis bandwidth: 1 GHz max. (MS2850A)
125 MHz max. (MS2690A/MS2691A/MS2692A/MS2830A)● Versatile built-in measurement functions: Adjacent channel leakage power,
MS2850 A
Occupied bandwidth, Spectrum emission mask, Spurious emission, Frequency counter, etc● Measurement software options supporting modulation analysis: 5G (MS2690A/MS2691A/MS2692A/MS2850A), LTE/LTE-Advanced,
W-CDMA/HSPA/HSPA Evolution, TD-SCDMA, GSM/EDGE/EDGE Evolution, CDMA2000/1xEV-DO, WLAN, FM/øM/AM● Vector Signal Generator option supporting modulation signal output (MS2690A/MS2691A/MS2692A/MS2830A)

Bluetooth Test Set
MT 8852 B
Corresponds to the latest Bluetooth core specification v5 . 1● Measurements performed as defined in the Bluetooth RF test specification● Supports Angle of Arrival/Angle of Departure added with Bluetooth core specification v5.1
● Qualified by Bluetooth SIG for RF measurements
The MT8852B is the market leading RF measuring instrument for design proving andproduction test of a wide range of products that integrate Bluetooth technology, including;phones, headsets, computers, audio-visual and gaming products as well as modules.
In production, a single key press initiates a measurement script that tests a device.

9

Cell Master™
MT 8213 E

Cable & Antenna Analyzer: 2 MHz to6 GHz, Spectrum Analyzer: 9 kHz to 6 GHz

Compact Base Station Analyzer● 30 analyzers in one● Cable and antenna analyzer: 2 MHz to 6 GHz● Return loss, Cable loss, VSWR, Distance-to-Fault● Spectrum analyzer: 9 kHz to 6 GHz● Interference analyzer with interference mapping, GPS

MOBILE/WIRELESS COMMUNICATION MEASURING INSTRUMENTS
10This optimal combination of base station test capabilities in one handheld device eliminates theneed for several independent test instruments, thereby reducing the number of tools the user mustcarry and learn to operate. Whether it's sweeping cables, making power measurements, finding
interference, troubleshooting base station signal quality, or verifying backhaul performance, theMT8213E is the ideal all-in-one instrument to help keep your network up and running.

BTS Master™
MT 8220 T

Cable & Antenna Analyzer: 400 MHz to 6 GHz, Spectrum Analyzer: 150 kHz to 7 . 1 GHz, Power Meter: 10 MHz to 7 . 1 GHz

High-performance Handheld Base Station Analyzer● 2-port cable and antenna analyzer: 400 MHz to 6 GHz● Spectrum analyzer: 150 kHz to 7.1 GHz● Power meter: 10 MHz to 7.1 GHz● GPS receiver with antenna
The MT8220T is the essential multi-function instrument for senior wireless technicians and RF
engineers to accurately and quickly verify the installation and commissioning of base stations for
optimal wireless network performance and for the on-going maintenance and troubleshooting
to keep wireless network infrastructure fine-tuned. A standard three-year warranty demonstrates
world-class reliability and brings peace-of-mind to owning and using the MT8220T BTS Master.

PIM Master™
MW 82119 B

Passive Intermodulation (PIM) Analyzer with Site Master™ Cable & Antenna Analyzer Option

Battery-operated, High Power Portable, Passive Intermodulation Analyzer with Cable & Antenna Analyzer● Passive intermodulation (PIM) analyzer● PIM vs. Time, Swept PIM, Distance-to-PIM, Noise floor● Battery operated: >3 hour
● 20 to 46 dBm (0.1 Watt to 40 Watt)● Field-proven design: Rugged, compact, daylight viewable display
The MW82119B is a 40 Watt, battery-operated PIM analyzer featuring Site Master™ line sweep
capability. With the Site Master option included, the MW82119B is able to fully certify cable andantenna system performance, measuring PIM, Distance-to-PIM, Return Loss, VSWR, Cable Loss
and Distance-to-Fault with a single test instrument. MW82119B includes a large, outdoor viewabledisplay and intuitive user interface that is optimized for field conditions. MW82119B’s rugged
design and enhanced portability enables both PIM and line sweep testing at the “top-of-the-tower,”
helping operators to achieve maximize RF performance from their LTE Remote Radio Head (RRH)
installations.

Signal Analyzer
MS2690 A/MS2691 A/MS2692 A

50 Hz to 6 GHz/13 .5 GHz/ 26 . 5 GHz

Signal analyzer with excellent fundamental performance● Frequency coverage up to 6 GHz/13.5 GHz/26.5 GHz● Total level accuracy: ±0.3 dB (typ.)● Dynamic range: 177 dB, TOI: ≥ +22 dBm, DANL: –155 dBm/Hz● Signal Analyzer ■ Analysis bandwidth: 31.25 MHz (standard), 62.5 MHz/125 MHz (option)■ Modulation analysis software: 5G, LTE/LTE-Advanced, WLAN (802.11a/b/g/n/j/p/ac), etc.
■ Capture and Playback function● Vector Signal Generator ■ Level accuracy: ±0.3 dB (typ.) ■ BER function, Internal AWGN generator

Signal Analyzer
MS2850 A series (MS2850 A-047 / 046 )9 kHz to 32 GHz/ 44 . 5 GHz

Wideband signal analysis using excellent dynamic range and flatness● Excellent Flatness Performance
Amplitude flatness: ±1.2 dB (nom.), Phase flatness: 5°p-p (nom.)
Center Frequency: 28 GHz, at Center Frequency ±500 MHz● Wide Dynamic Range
Better than 140 dB@28 GHz
The difference between the ADC clipping level and DANL● Analysis Bandwidth: 255 MHz (Standard), 510 MHz (Option), 1 GHz (Option)● Measurement applications (Options): Modulation Analysis (5G, LTE, LTE-Advanced, W-CDMA, TD-SCDMA, GSM, Vector Modulation, etc.),
Phase Noise, Noise Figure, Noise Floor Reduction, etc.
The MS2850A-047/046 is a spectrum analyzer/signal analyzer with a maximum analysis bandwidth of 1 GHz and excellent flatness
performance. With this performance, the MS2850A supports high-accuracy amplitude and phase measurements for each signal in
wideband next-generation communications systems, such as 5G mobile and satellite. In addition to signal analysis, 1 GHz analysis
bandwidth can be utilized as a digitizer application that monitors multiple frequencies in satellite communication. The large amountsof digitized data captured can be transfered to an external PC 100 times faster than conventional method by External Interface for High
Speed Data Transfer PCIe/USB3.0 MS2850A-053/054 option.

Signal Analyzer
MS2840 A series (MS2840 A-040 / 041 )9 kHz to 3 .6 GHz/ 6 GHz

Top Class Phase Noise Performance at Middle-Price Range● Phase Noise: – 140 dBc/Hz@150 MHz, 10 kHz offset (MS2840A-066, meas.)–138 dBc/Hz@1 GHz, 10 kHz offset (MS2840A-066, meas.)–123 dBc/Hz@1 GHz, 10 kHz offset (Standard)● Analysis Bandwidth: 31.25 MHz (Standard), 125 MHz max. (Option)● Measurement applications (Options): P hase Noise, Noise Figure, Vector and Analog
Modulation Analysis, Noise Floor Reduction, Built-in
Vector/Analog Signal Generator, BER

SIGNAL ANALYZERS/SPECTRUM ANALYZERSThe MS269xA has the excellent total level accuracy and dynamic range. Not only can it capture wideband signals but FFT technology
supports multifunction signal analyses in both the time and frequency domains.

MS2840A-041

Installing the MS2840A-066 option in the MS2840A-040/041 supports excellent phase noise performance exceeding that of high-end
models. It offers high cost-performance in fields including development and manufacturing of narrowband wireless equipment andoscillators with built-in wireless, as well as wireless fundamental research. Additionally, it is ideal for substitute replacement of firstgeneration and earlier legacy high-end models. It has a built-in signal analyzer function with a wide 31.25 MHz analysis bandwidth
using FFT technology for versatile analyses in both the time and frequency domains, etc. Moreover, installing the internal vector signal
generator and analog signal generator options provides all-in-one support for TRx measurements of wireless equipment.

Signal Analyzer
MS2840 A series (MS2840 A-044 / 046 )9 kHz to 26 . 5 GHz/ 44 . 5 GHz

Excellent Phase Noise Performance Using New Synthesizer Design● Phase Noise: – 123 dBc/Hz@1 GHz, 10 kHz offset–100 dBc/Hz@79 GHz, 10 kHz offset (with high performance waveguidemixer, meas.)● Support high performance waveguide mixer (50 GHz to 90 GHz) or harmonic mixer(up to 325 GHz)● Analysis Bandwidth: 31.25 MHz (Standard), 125 MHz max. (Option)● Measurement applications (Options): Pulse Radar Analysis, Phase Noise, Noise Figure,
Vector and Analog Modulation Analysis, Noise Floor Reduction, BER

MS2840A-046

The MS2840A-044/046 is a spectrum analyzer offering top-class phase noise performance in a middle–price-range model. This excellent
phase noise performance supports measurement of wideband transmitters, such as VHF and UHF LMR/PMR, where the measurement
instrument performance is key to measurement of close-in spurious, as well as measurement of microwave wireless backhaul, satellite,
radar, etc. Connection to two available high-performance waveguide mixers covers both V-band (50 GHz to 75 GHz) and E-band (60 GHz
to 90 GHz) measurements with the highest phase noise performance.

11

Signal Analyzer
MS2830 A series (MS2830 A-040 / 041 / 043 )9 kHz to 3 . 6 GHz/ 6 GHz/ 13 . 5 GHz

Supports TRx evaluations of various wireless equipment plus
spurious measurements of narrowband wireless equipment● Total level accuracy: ±0.3 dB (typ.) (300 kHz to 4 GHz)
● SSB phase noise: – 109 dBc/Hz@500 MHz, 1 kHz offset*–118 dBc/Hz@500 MHz, 10 kHz offset*–133 dBc/Hz@500 MHz, 100 kHz offset**: Requires Low Phase Noise Performance MS2830A-066

MS2830 A-041

SIGNAL ANALYZERS/SPECTRUM ANALYZERS● Measurement applications (Options):
Modulation Analysis (LTE/LTE-Advanced, Analog Modulation, Vector Modulation, etc.), Noise Figure, Built-in Audio Analyzer, Built-in
Vector/Analog Signal Generator, BER, Internal Signal Generator Control Function, etc.
With support for various measurements, the high cost-performance of the MS2830A series makes it ideal for various applications.
As well as TRx tests of different wireless equipment including digital and analog LMR/PMR/Transceiver Modules, and cellular and WLAN, it also
supports spurious measurements of narrowband wireless equipment typified by LMR/PMR. And the built-in Noise Figure (NF) measurement
function plus the Internal Signal Generator Control Function (for evaluating filter and amplifier transmission characteristics) expand theapplication range even further. Finally, Anritsu’s unique Capture & Playback Function can regenerate wireless equipment Tx signals.

Signal Analyzer
MS2830 A Microwave series (MS2830 A-044 / 045 )9 kHz to 26 . 5 GHz/ 43 GHz

For the Development & Manufacturing of the Millimeter-Wave Wireless Transmitters.
Spectrum Analyzer + Signal Analyzer● Total level accuracy: ±0.3 dB (typ.) (300 kHz to 4 GHz)● Dynamic range*: 159 dB@25 GHz *: (TOI–DANL)TOI: +13 dBm@25 GHz
DANL: –146 dBm/Hz@25 GHz
MS2830 A-045● SSB phase noise: –115 dBc/Hz@500 MHz, 100 kHz offset● Measurement applications (Options): Modulation Analysis (LTE/LTE-Advanced, Vector Modulation, etc.), Noise Figure, BER, etc.
The MS2830A-044/045 spectrum analyzer has an upper frequency limit of 26.5 GHz/43 GHz, which can be extended to 325 GHz using thehigh-performance waveguide mixer and external mixer. It can be customized to support various measurement applications.
• Confirming microwave signal frequency, phase, amplitude, instantaneous spectrum fluctuations, etc., in signal analyzer mode• Measuring weak signals at microwave preamplifiers• Measuring true spurious of increasingly wideband mm-Wave communications equipment using high IF (1.875 GHz) and highperformance waveguide mixer

High Performance Waveguide Mixer
MA 2806 A/MA 2808 A

50 GHz to 75 GHz/60 GHz to 90 GHz

Spectrum Analysis of Increasingly Wideband mm-Wave Transmitters● Easy set-up with one coaxial cable connection to MS2850A/MS2840A/MS2830A signal analyzer● Wide dynamic range using excellent minimum Rx sensitivity and P1dB performance● High IF and PS Function (patent pending) eliminating Image response effects at wideband signal
MA 2808 A
measurement● High phase noise performance of –100 dBc/Hz@79 GHz with 10 kHz offset (meas.) at connection with MS2840A● Easy loading of conversion loss data from accessory USB memory stick into signal analyzer for reflection in the measurement values
The MA2806A/MA2808A is a high-performance waveguide mixer for connection to the MS2850A, MS2840A-044/046 and MS2830A-044/045.
With high dynamic range performance, it is ideal for evaluating the true spurious of increasingly wideband mm-Wave transmitters. Moreover,when used with the high IF of the MS2850A/MS2840A/MS2830A*, it not only supports image-response-free measurements, but can also be used
for spectrum mask measurements of wideband signals, such as wireless backhaul and automobile radar, over a wide measurement span. Using
the newly developed, patentpending, PS Function, supports measurements without image responses up to a measurement span of 7.5 GHz.
*: MS2850A/MS2840A: 1.8755 GHz, MS2830A: 1.875 GHz

Field Master Pro™
MS2090 A

9 kHz to 9 /14 /20 /26 .5 /32 /43 .5 /54 GHz

Compact and Ruggedized for Field Use● RTSA bandwidth: 20 MHz to 100 MHz● Gives field engineers and technicians unparalleled measurement accuracy previously reserved for
only benchtop instruments● 5G NR base station measurement validates the performance of the gNB base station withessential measurements that are in full compliance with 3GPP TS 38.104 V15● Fast sweep speeds, low distortion front-end, and spectrogram display help deploy new network
efficiently, ensuring spectrums are clear and validating that all legacy users have stopped all transmission● A comprehensive range of transmitter measurements – including: harmonic, spurious, occupied bandwidth, channel power, andadjacent channel power, ensures conformity to regulatory requirements

12

The Field Master Pro MS2090A delivers the highest levels of RF performance available in a handheld, touchscreen spectrum analyzer,
with a displayed average noise level (DANL) of –164 dBm and Third Order Intercept (TOI) of +20 dBm (typical). This makes measurements
such as spectrum clearing, radio alignment, harmonic, and distortion even more accurate than previously possible. 100 MHz modulation
bandwidth coupled with best-in-class phase noise performance maximizes measurement precision provides confidence when testingtransmitter power and spurious.

Spectrum Master™ Ultraportable Spectrum Analyzer
MS2760 A/MS2762 A

9 kHz to 170 GHz

The Future of Performance and Affordability● mmWave capabilities for 5G, wireless backhaul, 802.11ad, satcom, and more● Ultraportable form factor enables measurements right at the device under test
● Measure: channel power, adjacent channel power, occupied bandwidth● Patented NLTL technology provides >100 dB dynamic range● –127 dBm DANL to 110 GHz● Up to 6 traces, 3 trace detectors, and 12 markers● Dynamic Range: >103 dB from 6.15 GHz up to 70 GHz

Spectrum Master™
MS2711 E/MS2712 E/MS2713 E

9 kHz to 3 GHz/4 GHz/ 6 GHz

Compact Handheld Spectrum Analyzer● Spectrum analyzer: 9 kHz to 6 GHz● Interference analyzer with interference mapping● High accuracy power meter, 2-port transmission measurements (MS2712E/MS2713E only)● Coverage mapping, Channel scanner, GPS, AM/FM/PM analyzer (No coverage mapping for MS2711E)● 3GPP, 3GPP2, WiMAX, ISDB-T, DVB-T/H signal analyzers (MS2712E/MS2713E only)● Tracking generator: 500 kHz to 4 GHz

MS2711 E

Regulatory requirements are growing. You're under increasing pressure to cut costs. And improving system uptime is always a toppriority. The MS271xE helps you do all of this and more. Whether you are performing complex interference analysis or assessing signal
quality, the MS271xE delivers the ease of use, rich functionality, and best-in-class price/performance you've come to expect from Anritsu.
Designed to handle the most punishing field conditions, the MS271xE allows you to monitor, locate, identify, and analyze a broad range ofcellular, 2G/3G/4G, land mobile radio, Wi-Fi, and broadcast signals. With a rich array of configuration options, the multifunctional MS271xE
eliminates the need for you to learn and carry multiple instruments when locating and identifying signals over wide frequency ranges.

Spectrum Master™
MS2720 T

9 kHz to 9 GHz/13 GHz/20 GHz/32 GHz/ 43 GHz

High-performance Handheld Spectrum Analyzer● Frequency coverage: 9 kHz to 43 GHz● Broadband preamplifiers over the whole frequency range for increased sensitivity approx. 17 dB● Three sweep modes: Improved sweep speed, up to 100 times faster● Resolution and video bandwidths from 1 Hz to 10 MHz● New triggering choices, including hysteresis, hold-off, and delay● More zero-span capabilities including 10 MHz RBW & VBW● Enhanced spectrum analyzer touch-screen GUI, including large marker display choice● Choice of display options for readability: normal, black on white, night vision, color on white, or high contrast● On-screen interference mapping as part of the interference analysis option

SIGNAL ANALYZERS/SPECTRUM ANALYZERSOur MS2760A Spectrum Master – Ultraportable Spectrum Analyzer utilizes Anritsu’s patented ShockLine non-linear transmission
line (NLTL) technology, the MS2760A shatters the cost, size, and performance barriers associated with traditional large form factor
instruments to more efficiently advance technology development. It is truly pocket-sized, but big on performance with industry leading
dynamic range, sweep speed, and amplitude accuracy.

Anritsu's spectrum analyzers, Spectrum Master MS2720T represents the company's highest performance handheld spectrum analyzer.
Exciting new features and options bring more value and speed to the user.

Remote Spectrum Monitor
MS27101 A/MS27102 A/MS27103 A

9 kHz to 6 GHz

For Remote RF Signal Monitoring● Frequency coverage: 9 kHz to 6 GHz● Sweep speed up to 24 GHz/s● Integrated web server to view, control and conduct measurements via web browser● Watchdog timer to insure long-term stability for remotely deployed monitors● Low spurious levels for accurate signal discovery● 20 MHz instantaneous FFT bandwidth
Our three models of remote spectrum monitoring products are designed to both mitigate
interference problems and to identify illegal or unlicensed signal activity. The MS27101A is housed
in a 1/2 rack enclosure with 1U height, designed exclusively for indoor applications. MS27102A is an
IP67 rated device which operates outdoors, with the ability to be mounted on poles or walls (using
the included mounting bracket). The MS27103A is a multi-port spectrum monitor (12 RF In ports or
optionally 24 RF In ports) which is ideal for cellular, DAS and other applications requiring the use ofmultiple antennas.

MS27101 A

MS27102 A

MS27103 A

13

ShockLine™ 1-Port VNA
MS46121 B series

40 MHz to 4 GHz/150 kHz to 6 GHz

Ideal for Testing Multiple 1-port Devices in Parallel for Improved Test Productivity and Throughput● 1-port VNA with frequency options from 150 kHz to 6 GHz● External PC control enables control of multiple MS46121B in parallel for excellent multisite throughput● Very compact package allows for direct connection to the DUT● Scalar transmission measurements in a (1-to-1) or (1-to-n) configuration● No onboard data storage eliminates the need for data purging in secure applications● Standard bandpass time domain with time gating grants easier and faster fault identification● A common GUI interface within the ShockLine family reduces switching costs between models

VECTOR NETWORK ANALYZERSThe MS46121B is a series of two PC-controlled 1-port ShockLine Vector Network Analyzers with frequency ranges of 40 MHz to 4 GHz
and 150 kHz to 6 GHz. The MS46121B provides performance and accuracy for your 1-port measurements in a low cost and spacesaving solution that is small enough to directly connect to the device under test. All the members of the MS46121B are aimed at RF andmicrowave applications in manufacturing, engineering and education. The two MS46121B options both come with 100 ms/point sweep
speeds and a measurement accuracy of ±0.5 dB (–6 dB offset, typ.), making them suitable for your passive device test applications.

ShockLine™ Compact VNA
MS46122 B series

1 MHz to 8 GHz/ 20 GHz/ 43 . 5 GHz

Low-cost Series of 1 U High, 2 -port Compact Vector Network Analyzers● World's first series of compact VNAs to 43.5 GHz for cost-effective measurements● PC control takes advantage of external computer processing power and functionality● Compact 1U high package for efficient use of bench and rack space● No onboard data storage eliminates the need for data purging in secure applications● Time domain with time gating option grants easier and faster fault identification
The MS46122B is a series of three PC-controlled Compact ShockLine Vector Network Analyzers with a frequency range from 1 MHz
to 8 GHz/20 GHz/43.5 GHz. The series benefits from patented ShockLine VNA-on-chip technology, which simplifies the internal VNA
architecture at high frequencies, reduces instrument cost, and enhances accuracy and measurement repeatability.

ShockLine™ Economy VNA
MS46322 B series

1 MHz to 8 GHz/ 20 GHz/ 43 . 5 GHz

Low-cost Series of 2 U High, 2 -port Economy Vector Network Analyzers● Ideal for testing RF and microwave devices● Fast sweep speed and wide dynamic range minimize test times and maximize throughput● Excellent corrected directivity allows for less measurement uncertainty● Time domain with time gating option grants easier and faster fault identification● The LAN interface for remote control is more robust than USB and faster than GPIB
● A common GUI and SCPI interface within the ShockLine family● USB ports allow for easy connection to user-provided monitor, keyboard, and mouse● The small 2U packages allows for the efficient use of rack space
The MS46322B is a series of Economy ShockLine Vector Network Analyzers with frequency range from 1 MHz to 8, 20 and 43.5 GHz.
It is based on patented ShockLine VNA-on-chip technology, which simplifies the internal VNA architecture at high frequencies, reduces
instrument cost, and enhances accuracy and measurement repeatability.

ShockLine™ 2-Port Performance VNA
MS46522 B/MS46524 B series

50 kHz to 8 . 5 GHz/ 20 GHz/ 43 . 5 GHz (55 GHz to 92 GHz)

High-performance, 3 U High, 2 -port VNA Available in a 50 kHz to 43 .5 GHz Frequency Range● High output power allows measurement of high attenuation devices (MS46522B)● Industry leading dynamic range enables measurement of very low reflection artifacts● Excellent corrected directivity minimizes measurement uncertainty● SmartCal™ automatic calibration unit reduces calibration and setup time● Time domain with time gating option grants easier and faster fault identification● Modern LAN interface for remote control is faster than GPIB
● A common GUI and SCPI interface within the ShockLine family● E-band VNA ■ Extended frequency range covering E-band and major parts of V-band■ Full-assembled test system eliminates setup errors and increases reliability■ Tethered modules connect directly to the DUT increasing measurement stability● Simple signal integrity testing of passive multi-port and differential devices● The compact 3U high chassis allows for the efficient use of rack space
The MS46522B/MS46524B is a series of 2 and 4-port Performance ShockLine Vector Network
Analyzers. Delivering an unprecedented level of value and performance, including best-inclass dynamic range, the Performance series lowers cost-of-test and speeds time to market
in numerous testing applications up to 92 GHz. These applications include designing andmanufacturing mobile network equipment, mobile devices, automotive cables, high-speed
data interconnects and system integration components.

14

MS46522 B

MS46524 B

VectorStar™ Microwave VNA
MS4640 B

10 MHz to 20 GHz/ 40 GHz/70 GHz

Premium Family of RF to Microwave and Millimeter-wave Vector Network Analyzers● Broadest frequency span from a single coaxial test port covering 10 MHz to 70 GHz in a single
instrument and 70 kHz to 145 GHz in the broadband configuration. Extendable to 1.1 THz● IMDView™ software coupled with the internal combiner option offers the ability to switchfrom S-parameters to IMD measurements in a single connection● Highest performance pulse measurements — PulseView™ offers 2.5 ns pulse resolution with
100 dB dynamic range● 4-port single-ended or balanced measurements using DifferentialView™ analysis● Superior dynamic range: up to 142 dB

VectorStar™ Broadband VNA
ME 7838 A/A 4 /E/D series

70 kHz to 110 GHz/145 GHz

High-performance, Broadband Network Analysis Solutions● The ME7838A or E version can easily be upgraded to 145 GHz● All versions may be configured to include banded mm-wave modules up to 1.1 THz● Industry-best calibration and measurement stability: 0.1 dB vs. 0.6 dB over 24 hours● All versions support the 3744x-Rx receiver for noise figure measurements to 125 GHz● Compact, lightweight mm-wave modules (0.6 lb. vs. 7+ lbs. and 1/50 the volume) offer
low cost installation on smaller probe stations
The ME7838 series system provides high-performance in a compact mm-wave modulewith industry-best calibration stability. While other broadband systems continue toprovide raw performance with negative directivity in critical frequency bands, theME7838 series is the only broadband system with positive raw directivity in all bands.
The result is better calibration stability and better measurement stability withsignificantly longer time between calibrations for accurate measurements and improved
productivity. The ME7838D takes high-performance broadband measurements to a newlevel with the addition of the mm-wave module MA25300A. The MA25300A adds the next
waveguide band and combines it with the Anritsu developed 0.8 mm coaxial connector
offering the world’s first broadband VNA operating beyond the limits of W-band in a single sweep.

ME 7838 A

VECTOR NETWORK ANALYZERSThe VectorStar™ family is Anritsu's premium VNA line, providing the highest overall performance on a modern platform.
The MS4640B offers the broadest coverage in a single instrument, 70 kHz to 70 GHz. The additional two decades at the low-end are evenmore impressive than the guaranteed 70 GHz coverage on the high-end.

VNA Master™
MS202 xC/MS203 xC series

5 kHz to 6 GHz/15 GHz/20 GHz

The Ultimate Handheld Vector Network + Spectrum Analyzer for Cable, Antenna and Signal Analysis Anytime, Anywhere
VNA Master (MS202xC/MS203xC)● True 2-path 2-port fully-reversing VNA● Ultra-fast 350 μs/data point sweep speed● 12-term error correction algorithm● Vector voltmeter and time domain option● User-defined quad display for viewing all 4 S-parameters

Spectrum Analyzer: 9 kHz to 9 GHz/15 GHz/20 GHz (MS203xC)● 	Detectors: Peak, Negative, Sample, Quasi-peak, and True-RMS● 	Markers: 6, each with a delta marker, or 1 reference with 6 deltas● 	Built-in pre-selector for eliminating spurious in displays

The MS202xC/MS203xC series offers the industry's first 12-term error correction algorithm in a handheld
VNA. With a typical measurement speed of 350 μs/point, it is ideally suited for tuning filters in the field
where multiple S-parameters often interact during tuning. Using a 3-receiver architecture, the MS202xC/
MS203xC can measure and display all 2-port S-parameters at once with a one-time connection to the DUT.
It specifically addresses complex cable and antenna measurement needs in the field with accurate, vector
corrected 2-port magnitude, phase, and Distance-to-Fault measurements.
MS2026 C

Site Master™
S331 P

Compact handheld cable & antenna analyzer: 150 kHz to 6 . 0 GHz

Ultraportable Cable & Antenna Analyzer Featuring Classic and Advanced Modes● Smallest, lightest, and fastest Site Master™
● Direct connection to DUTs eliminating the need for phase stable cables● Powered through USB interface (No battery required)● Rugged and reliable● Impact, dust and splash resistant● Compatible with Anritsu Software tools including easyTest™ and Skybridge™ Tools
The S331P is an ultraportable version of the industry-leading Site Master Series of Cable and Antenna Analyzers. It is the smallest,lightest, fastest, and most cost effective instrument in the Site Master family. No battery is required, since the USB port of Windows
7, 8 & 10 tablet devices, laptops or desktop PCs power it. Available with two frequency ranges starting from 150 kHz up to 4 GHz and
150 kHz to 6 GHz, it is the only small headless Site Master product capable of measurements down to 150 kHz for low frequency radio
communications applications and up to 6 GHz for higher frequency applications like LTE-U in the 5 GHz unlicensed spectrum.

15

Site Master™
S331 L2 MHz to 4 GHz, Power Meter: 50 MHz to 4 GHz

Handheld Cable & Antenna Analyzer Featuring Classic and Advanced Modes● 2 MHz to 4 GHz handheld cable and antenna analyzer, impact, dust, and splash resistant● More than 8 hours of continuous battery operation● Standard built-in InstaCal™ module and power meter● FlexCal™ maintains calibration with frequency changes● Familiar S331D-like classic mode and S331E-like advanced mode
● Built-in one button help function● 800 × 480 7-inch TFT touch-screen display and multiple USB ports

VECTOR NETWORK ANALYZERSThe S331L is an all-inclusive 1-port cable and antenna analyzer covering the 2 MHz to 4 GHz range, with a built-in InstaCal module anda built-in power meter. Standard measurements include: return loss, VSWR, cable loss, smith chart (50Ω/75Ω selectable), 1-port phase,
distance-to-fault, return loss, distance-to-fault VSWR, RF power (50 MHz to 4 GHz), and VIP mode, optical connector inspection with IEC
61300-3-35 based pass/fail standard (requires USB video inspection probe G0306A, sold separately).

Site Master™
S331 E/S332 E/S361 E/S362 E

2 MHz to 4 GHz/6 GHz, Spectrum Analyzer: 9 kHz to 4 GHz/6 GHz, Cable and Antenna Analyzer: 9 kHz to 4 GHz/6 GHz

Compact Handheld Cable and Antenna Analyzers with Spectrum Analyzer● Cable and Antenna Analyzer: 2 MHz to 4 GHz● Return Loss, VSWR, Cable Loss, Distance-To-Fault, Smith Chart, 1-Port Phase● Field-proven design: Four-hour battery life, rugged, compact, lightweight, daylight viewable display● USB connectivity, built-in touch screen keyboard● Intuitive menu-driven touch screen user interface● Standard three-year warranty (battery one-year warranty)
The Site Master is the preferred cable and antenna analyzer of wireless service providers,
contractors and installers. It is the most integrated cable and antenna analyzer in the world.
Our compact handheld cable and antenna analyzer with spectrum analyzer is a sleek, compact
instrument that’s less than 6 lbs.

S331 ELMR Master™
S412 E500 kHz to 1 . 6 GHz/ 6 GHz, Spectrum Analyzer: 9 kHz to 1 . 6 GHz

Land Mobile Radio Modulation Analyzer and Signal Generator, Vector Network Analyzer, Spectrum Analyzer● Cable and antenna analyzer: 500 kHz to 1.6 GHz, optional to 6 GHz● Return loss, VSWR, Insertion loss, S11/S21, DTF● Spectrum analyzer: 9 kHz to 1.6 GHz, optional to 6 GHz● NBFM signal analyzer with coverage mapping● LMR signal analyzers with coverage mapping: P25, P25 phase 2, NXDN, DMR, PTC, TETRA● Broadband signal analyzers: LTE, WiMAX● Interference analyzer with interference mapping and support for Handheld InterferenceHunter MA2700A
The S412E is the ideal instrument for Land Mobile Radio (LMR), Professional Mobile Radio (PMR)
technicians and engineers engaged in field testing the RF performance of NBFM, P25, P25 Phase 2(TDMA), NXDN, ETSI DMR, MotoTRBO, ETSI TETRA, and LTE for commercial, public safety, maritime,and critical infrastructure radio systems. In addition the S412E offers support for USA class 1 railway
Positive Train Control systems.

Microwave Site Master™
S820 E

1 MHz to 8 GHz/ 14 GHz/ 20 GHz/ 30 GHz/ 40 GHz

Cable & Antenna Analyzer● Frequency range: 1 MHz to 8, 14, 20, 30, 40 GHz● VNA mode (option) offers fully reversing 4S-parameter measurement capabilities● VVM mode (option) with standard A/B and B/A ratio capability● 110 dB of dynamic range from 20 MHz to 40 GHz● 650 μs/data point for fast field measurements● Advanced and classic mode GUI (i.e. S810D/S820D)● Coaxial and waveguide measurement supported
The S820E family with frequency options covering 1 MHz to 8, 14, 20, 30, and 40 GHz, is the world’s most advanced Site Master everdeveloped. Available Vector Network Analyzer (VNA) and Vector Voltmeter (VVM) options allow users to easily expand the S820E’s
versatility at any time. Adding options to the S820E is hassle free, simply purchase the desired option(s) and install the option activation
key(s) provided by Anritsu. No need to send the instrument into a service department because all hardware and calibrations required are
already built into the S820E before it leaves the factory.

16

RF/Microwave Signal Generator
MG3690 C series

0 .1 Hz to 70 GHz/500 GHz

The Ideal Microwave Signal Generators● Industry best broad frequency coverage● Industry best phase noise● Industry best pulse modulation● Fast switching speed● Full analog modulation● High output power● Highly configurable● Highly reliable

The MG3690C series of broadband signal generators covers audio, HF, VHF, UHF, RF and microwave frequencies from 0.1 Hz to 70 GHz
in single coaxial output and up to 500 GHz or beyond, with external multipliers. In addition, it is easy to operate either through intuitive
front panel controls or remotely via GPIB or Ethernet connectivity. Thus, the MG3690C series is an ideal signal source for both RF andmicrowave requirements, fully configurable for simple to high-performance applications.

MG3710 E

100 kHz to 2 . 7 GHz/ 4 GHz/6 GHz

Multi-band, Multi-system, Multi-channel Cut Costs for New Wireless Tests● Generate and output modulation signals of various communication systems such as
5G NR (sub-6 GHz), LTE-Advanced FDD/TDD (Optional)● RF modulation bandwidth 160 MHz*/120 MHz● ACLR: –71 dBc (W-CDMA, TestModel1, 64DPCH, 2 GHz)
● SSB phase noise: <–131 dBc/Hz (typ.) (1 GHz, 20 kHz offset, CW)● Pre-installed key waveform patterns● Waveform addition function:
Adds and outputs two signals, such as wanted signal + interference signal or wanted
signal + AWGN● One unit supports two RF outputs max. Ideal for multi-system evaluations.
● Supports BER test function

SIGNAL GENERATORS

Vector Signal Generator*: When using MX370111A/MX370111A-002

The MG3710E is a Vector Signal Generator with 6 GHz upper frequency limit and 160 MHz*/120 MHz wide RF modulation baseband
generator. The excellent signal generator ACLR and SSB phase noise reduces the effect on wideband and narrow-band measurements
to improve test margins and yields. It outputs various wireless systems signals such as 5G, LTE, WLAN and narrowband communications(PMR/LMR).

Analog Signal Generator
MG3740 A

100 kHz to 2 . 7 GHz/ 4 GHz/6 GHz

Versatile Modulation Functions, Excellent Expandability● Built-in AM/FM/øM/Pulse modulation function (standard)● Additional analog modulation input (option)
AM + FM, AM + øM, Internal 1 + Internal 2, Internal + External*: FM + øM does not support.

● Dual RF Outputs (option)
One unit supports two RF outputs (1st RF/2nd RF) max.
● Narrowband digital modulation function (option)
RF modulation bandwidth: 2 MHz● BER test function (option), Input bit rate: 100 bps to 40 Mbps
● USB power sensors (sold separately)
The MG3740A has excellent RF specifications, including SSB phase noise, output level, etc., and versatile modulation functions (AM/FM/
øM/Pulse). Moreover, the MG3740A supports additional analog modulation by external signal input, dual RF outputs, narrowband digitalmodulation function for private mobile radio (PMR), BER test function and USB power sensors.

17

Wideband Peak Power Meter
ML249 xA series

100 kHz to 65 GHz

World’s most comprehensive range of power meters● Ideal for measuring radar and communication signals like WiMAX, W-CDMA, WLAN, GSM etc.
● CW and average power measurements as low as –60 dBm● 20 MHz video bandwidth● Sampling rate of 64 MS/s with ML2490A series power meters
The ML2490A series are designed for use with the Wideband Peak Power Sensors MA249xA.
These power meters and sensors provide peak, crest factor, average power, rise time, fall time,
maximum power, minimum power and statistical data of wideband signals.

RF MICROWAVE MEASURING INSTRUMENTS

Power Meter
ML2437 A/ML2438 A
For Measuring Wide Dynamic Range Power● Speed accuracy and flexibility in a low cost package● Portable, rugged, and splash resistant● Optional Ni-MH battery, providing six hours continuous operation● Measure and transfer a high speed burst of 200 data points using profile operating mode withsampling rates of 35k per second● With 99.9% emulation of older meters, the ML2430A series improves ATE system productivity
Typical test system speed improvement is 2 to 10 times faster system speed● Single input (ML2437A), Dual input (ML2438A)
The ML2437A/ML2438A combines the advantages of thermal meter accuracy, diode meter
speed, and peak power meter display graphics. The result is a single instrument that samples atmore than 35k per second and achieves 90 dB dynamic range with a single sensor.
This meter includes graphics display capability as a standard feature. The ruggedized housing
and optional high-capacity Ni-MH battery bring laboratory quality accuracy to field service applications.

Power Master™

only MS24507A

MA 24507 A/MA 24510 A
Frequency Selectable mmWave Power Analyzer● Low power capability to measure signals as low as –90 dBm● Excellent for over-the-air testing, especially with mmWave signals that have high propagation loss
● User settings to control measurement speeds and noise floor● Channel Monitor mode in PowerXpert for monitoring up to six frequency channels at once
● Power Hunter mode in PowerXpert for searching up to six signals within a frequency range● Mounting holes for direct mounting to connect probes for over-the-air or on-wafer testingPower Master is an ultraportable, USB-powered mmWave power analyzer that enables simple,
numeric, frequency-based measurement of RF power from 9 kHz to 110 GHz and as low as –90
dBm. Traditional power meters are broadband and have limited power ranges, so engineers andtechnicians are using spectrum analyzers that include many unneeded features, cost hundredsof thousands of dollars, and take up half the test bench just to make simple, frequency-based RF
amplitude measurements. The Power Master MA24507A enables those measurements in a USBpowered device slightly bigger than a smartphone and at a fraction of the price of a spectrum analyzer.

Inline Peak Power Sensor
MA 24105 A

350 MHz to 4 GHz

A Standalone, Compact, and Highly Accurate Bi-directional Inline Peak Power Sensor● Broad frequency range: 350 MHz to 4 GHz● Covers all major cellular and communication bands, such as GSM/EDGE, CDMA/EV-DO, W-CDMA/HSDPA,
WiMAX and TD-SCDMA● Forward and reverse measurements● Widest dynamic range inline power sensor in its class● True-RMS measurements to 150 W● Standalone, Low cost, Plug and Play device

18

The MA24105A is designed to take accurate average power measurements over 2 mW to 150 W, from 350 MHz to 4 GHz. The sensor
employs a "dual path" architecture that enables True-RMS measurements over the entire frequency and dynamic range allowing users tomeasure CW, multi-tone and digitally modulated signals such as GSM/EDGE, CDMA/EV-DO, W-CDMA/HSDPA, WiMAX, and TD-SCDMA.
The forward direction path also includes a 4 MHz bandwidth channel that has peak and comparator/integrator circuits that add
measurement functions such as PEP power, crest factor, CCDF, and burst average power. Another detection circuit on the reverse direction
adds reverse power measurement capabilities including reverse power, reflection coefficient, return-loss, and SWR. The presence of a
micro-controller along with signal conditioning circuitry, ADC, and power supply in the sensor makes it a complete miniature power meter.

USB Power Sensor
MA 24106 A

True-RMS, 50 MHz to 6 GHz

Handy, Highly Accurate and Reliable USB Sensor for RF Power Measurement● True-RMS measurements over 63 dB dynamic range enables accurate CW andmodulated power measurements● Ready for use in a wide variety of applications, including installation and maintenance ofbase stations, testing of 3G and 4G devices, cell phones and general purpose RF devices● High damage power levels and ESD protection circuitry showcases ruggedness
and reliability● Low power consumption (100 mA, typ.) extends laptop battery life
● Worldwide calibration and service centers ensure reduced downtime and local support

Microwave USB Power Sensor
MA 24108 A/MA 24118 A/MA 24126 A

10 MHz to 8 GHz/ 18 GHz/ 26 GHz

Low-cost, Compact, and Highly Accurate Power Sensors for RF and Microwave Applications● Broad frequency range: 10 MHz to 8 GHz (MA24108A)/18 GHz (MA24118A)/
26 GHz (MA24126A)● True-RMS measurements● NIST traceable calibration● 	Built-in internal and external trigger (only used with PC)● High power handling (+33 dBm)
● 1 mW calibration need eliminated● Silicone protective covering for additional field durability
The MA24108A/MA24118A/MA24126A are designed to provide accurate average power measurements from 10 MHz to 8 GHz/18 GHz/26
GHz over 60 dB of dynamic range. These sensors employ a "dual path" architecture that provides (similar to thermal sensor) True-RMS
measurements over the entire frequency and dynamic range, enabling users to make highly accurate average power measurements for
CW, multi-tone, and digitally modulated signal up to 26 GHz. The sensors have internal and external triggering capability that facilitates
individual slot power measurements of TDMA waveforms as well as burst power measurements of periodic and non-periodic waveforms.

Microwave Universal USB Power Sensor
MA 24208 A/MA 24218 A

10 MHz to 8 GHz/18 GHz

Low-cost, Compact, and Highly Accurate Power Sensors for RF and Microwave Applications

RF MICROWAVE MEASURING INSTRUMENTS

The MA24106A is a USB power sensor that eliminates the need of a traditional power meter. It is a highly accurate, standalone
instrument that communicates with a PC via USB. The power measurement capability of MA24106A is intended to mimic that of a
traditional thermal (thermo-electric) power sensor with a wider dynamic range.

● Frequency range: 10 MHz to 8 GHz (MA24208A)/18 GHz (MA24218A)● Power measurement range: –60 to +20 dBm● True-RMS measurements – modulation independent● Fast measurement speed: >1,600 readings/s continuous, >11,000 readings/s buffered● Able to accept high power levels before being damaged: +30 dBm (CW), +34 dBm (peak <10 µs)● No zero required● NIST traceable calibrations (MA24218A)
The MA24208A and MA24218A are designed to provide fast, accurate average power measurements from 10 MHz to 8 GHz/18 GHz over
80 dB of dynamic range. These sensors employ a patented "triple path" architecture that provides True-RMS measurements (similar tothermal sensors) over the entire frequency and dynamic range, enabling users to make highly accurate average power measurements
for CW, multi-tone, and digitally modulated signal up to 18 GHz. These sensors employ high-performance digital processing that enables
best-in-class measurements speeds, including >1,600 continuous power readings/s continuous and >11,000 buffered readings/s.

19

Speciﬁcations are subject to change without notice.

• United States
Anritsu Americas Sales Company
450 Century Parkway, Suite 190, Allen,
TX 75013, U.S.A.
Phone: +1-800-Anritsu (1-800-267-4878)• Canada
Anritsu Electronics Ltd.

700 Silver Seven Road, Suite 120, Kanata,
Ontario K2V 1C3, Canada
Phone: +1-613-591-2003
Fax: +1-613-591-1006• Brazil
Anritsu Eletronica Ltda.

Praça Amadeu Amaral, 27 - 1 Andar
01327-010 - Bela Vista - Sao Paulo - SP, Brazil
Phone: +55-11-3283-2511
Fax: +55-11-3288-6940• Mexico
Anritsu Company, S.A. de C.V.

Blvd Miguel de Cervantes Saavedra #169 Piso 1, Col. Granada
Mexico, Ciudad de Mexico, 11520, MEXICO
Phone: +52-55-4169-7104• United Kingdom
Anritsu EMEA Ltd.

200 Capability Green, Luton, Bedfordshire, LU1 3LU, U.K.
Phone: +44-1582-433200
Fax: +44-1582-731303• France
Anritsu S.A.

12 avenue du Québec, Bâtiment Iris 1- Silic 612,
91140 VILLEBON SUR YVETTE, France
Phone: +33-1-60-92-15-50
Fax: +33-1-64-46-10-65• Germany
Anritsu GmbH

Nemetschek Haus, Konrad-Zuse-Platz 1
81829 München, Germany
Phone: +49-89-442308-0
Fax: +49-89-442308-55• Italy
Anritsu S.r.l.

Via Elio Vittorini 129, 00144 Roma, Italy
Phone: +39-6-509-9711
Fax: +39-6-502-2425• SwedenAnritsu AB

Isafjordsgatan 32C, 164 40 KISTA, Sweden
Phone: +46-8-534-707-00• Finland
Anritsu AB

Teknobulevardi 3-5, FI-01530 VANTAA, Finland
Phone: +358-20-741-8100
Fax: +358-20-741-8111• DenmarkAnritsu A/S

c/o Regus Fairway, Arne Jacobsens Allé 7, 5th ﬂoor
2300 Copenhagen S, Denmark
Phone: +45-7211-2200• Russia
Anritsu EMEA Ltd.
Representation Oﬃce in Russia

Tverskaya str. 16/2, bld. 1, 7th ﬂoor.
Moscow, 125009, Russia
Phone: +7-495-363-1694
Fax: +7-495-935-8962• Spain
Anritsu EMEA Ltd.
Representation Oﬃce in Spain

Paseo de la Castellana, 141. Planta 5, Ediﬁcio Cuzco IV
28046, Madrid, Spain
Phone: +34-91-572-6761• United Arab Emirates
Anritsu EMEA Ltd.
Dubai Liaison Oﬃce

• Singapore
Anritsu Pte. Ltd.

11 Chang Charn Road, #04-01, Shriro House
Singapore 159640
Phone: +65-6282-2400
Fax: +65-6282-2533• P.R. China (Shanghai)
Anritsu (China) Co., Ltd.

Room 2701-2705, Tower A,
New Caohejing International Business Center
No. 391 Gui Ping Road Shanghai, 200233, P.R. China
Phone: +86-21-6237-0898
Fax: +86-21-6237-0899• P.R. China (Hong Kong)
Anritsu Company Ltd.

Unit 1006-7, 10/F., Greenﬁeld Tower, Concordia Plaza,No. 1 Science Museum Road, Tsim Sha Tsui East,
Kowloon, Hong Kong, P.R. China
Phone: +852-2301-4980
Fax: +852-2301-3545• JapanAnritsu Corporation

8-5, Tamura-cho, Atsugi-shi, Kanagawa, 243-0016 Japan
Phone: +81-46-296-6509
Fax: +81-46-225-8352• Korea
Anritsu Corporation, Ltd.

5FL, 235 Pangyoyeok-ro, Bundang-gu, Seongnam-si,
Gyeonggi-do, 13494 Korea
Phone: +82-31-696-7750
Fax: +82-31-696-7751• Australia
Anritsu Pty. Ltd.

902, Aurora Tower,
P O Box: 500311- Dubai Internet City
Dubai, United Arab Emirates
Phone: +971-4-3758479
Fax: +971-4-4249036• India
Anritsu India Private Limited

6th Floor, Indiqube ETA, No.38/4, Adjacent to EMC2,
Doddanekundi, Outer Ring Road, Bengaluru – 560048, India
Phone: +91-80-6728-1300
Fax: +91-80-6728-1301

Unit 20, 21-35 Ricketts Road,
Mount Waverley, Victoria 3149, Australia
Phone: +61-3-9558-8177
Fax: +61-3-9558-8255• Taiwan
Anritsu Company Inc.

7F, No. 316, Sec. 1, NeiHu Rd., Taipei 114, Taiwan
Phone: +886-2-8751-1816
Fax: +886-2-8751-1817

1911

Printed in Japan 4/FEB/2020 13815ddcw Catalog No. 2020EMI_Digest-E-A-1-(1.00)