MODEL 8711-01
ACCELEROMETERSPECIFICATIONS

DIMENSIONS

Industrial IEPE Accelerometer
Low Cost, Quick Delivery-55°C to +125°C Operating Range
Top Exit Connector

The Model 8711-01 are internally shielded rugged IEPE
accelerometers designed for industrial condition
monitoring. The accelerometers are available in ±5g to±80g dynamic ranges and have a flat bandwidth up togreater than 10kHz. The model 8711-01 accelerometerfeatures a top exit MIL-C-5015 connector and is designedto operate in temperature ranges from -55°C to +125°C.

FEATURES

±5g, ±10g, ±20g & ±80g Dynamic Ranges
Wide bandwidth up to >10kHz
Case Isolated, Internally Shielded
Hermetically Sealed, Welded
Annular Shear Mode
Reverse Wiring Protection
Stable Temperature Response

APPLICATIONSSENSOR SOLUTIONS /// Model 8711-01 Rev A

General Purpose
Machine Monitoring
Industrial Applications
Harsh Environments
Gearbox Monitoring

07/2017

Page 1

MODEL 8711-01 ACCELEROMETERPERFORMANCE SPECIFICATIONS
All values are typical at +24°C, 80Hz and 4mA excitation unless otherwise stated. Measurement Specialties reserves the right to update
and change these specifications without notice. Standard product parameters are described in PSC-1003 for Plug & Play AC
Accelerometers.

1Parameters
DYNAMIC
Range (g)Sensitivity (mV/g)
Frequency Response (Hz)Frequency Response (Hz)Natural Frequency (Hz)Non-Linearity (%FSO)
Transverse Sensitivity (%)
Shock Limit (g)Residual Noise (g RMS)
Weight (grams)±510000.3-4000
0.1-6000
20000±1<520000.00003
105

ELECTRICAL
Compliance Voltage (Vdc)
Excitation Current (mA)
Bias Voltage (Vdc)
Output Impedance (Ω)
Insulation Resistance (MΩ)
Warm-up Time (sec)Grounding18 to 30
2 to 1010 to 13<100>100<5Case Isolated, Internally Shielded

ENVIRONMENTAL
Temperature Response (%)
Operating Temperature (°C)
Storage Temperature (°C)
Humidity

See typical temperature response on following page
-55 to +125
-55 to +125
Hermetically Sealed, IP67

PHYSICAL
Sensing Element
Case Material
Mounting TorqueCeramic (shear mode)
Stainless Steel24 lb-in (2.7 N-m)

Notes
±105000.3-4000
0.1-600020000±1<52000
0.00005105±20250
1-70000.3-10000
32000±1<550000.00010
88±80
1001-7000
0.3-10000
32000±1<550000.00015
88±10%±5%±3dB

See Note 1
Room Temperature@100Vdc

For full dynamic range a minimum compliance voltage of 24Vdc is recommended

Calibration supplied:

CS-SENS-0100

NIST Traceable Amplitude Calibration at 80Hz

Supplied accessories:

AC-A03663¼-28 to ¼-28 mounting stud

Optional accessories:

316A-XXX
318A-XXX
AC-D03664
AC-D03665

Cable Assembly, Straight (XXX designates length in inches, 10ft standard)
Cable Assembly, 90 Degrees (XXX designates length in inches, 10ft standard)¼-28 to M5 mounting stud¼-28 to M6 mounting stud

The information in this sheet has been carefully reviewed and is believed to be accurate; however, no responsibility is assumed for inaccuracies. Furthermore, thisinformation does not convey to the purchaser of such devices any license under the patent rights to the manufacturer. Measurement Specialties, Inc. reserves theright to make changes without further notice to any product herein. Measurement Specialties, Inc. makes no warranty, representation or guarantee regarding thesuitability of its product for any particular purpose, nor does Measurement Specialties, Inc. assume any liability arising out of the application or use of any product orcircuit and specifically disclaims any and all liability, including without limitation consequential or incidental damages. Typical parameters can and do vary in different
applications. All operating parameters must be validated for each customer application by customer’s technical experts. Measurement Specialties, Inc. does notconvey any license under its patent rights nor the rights of others.

SENSOR SOLUTIONS /// Model 8711-01 Rev A

07/2017

Page 2

MODEL 8711-01 ACCELEROMETER

Typical Frequency Response40353025

Sensitivity Deviation (%)

20151050-5-10-15-20-25-30

Standard Option-35

LF Option-400110100

1000

10000

100000Frequency (Hz)SENSOR SOLUTIONS /// Model 8711-01 Rev A

07/2017

Page 3

MODEL 8711-01 ACCELEROMETER

ORDERING INFORMATIONPART NUMBERING

Model Number+Range

8711LF-01-GGG
ll_____ Range (080 is 80g)
l__________ Low frequency, required for 5g and 10g ranges
Standard model numbers:
8711LF-01-005
8711LF-01-010
8711-01-020
8711-01-080

TE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred to herein
might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard
Terms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuseof the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies

All Rights Reserved.

SENSOR SOLUTIONS /// Model 8711-01 Rev A

07/2017

Page 4