The Complete X-ray Inspection Solution
Quadra 7 Pro represents the cutting edge of X-ray inspection performance.
6.5 MP ultra high quality images are displayed at full one-to-one resolution over one single curved ultrawide
4K monitor, ensuring you see the most modern features sharply and in all lucidity.

www.nordson.com/TestInspect

Manual X-ray Inspection

Pioneering Software

Get working quickly with Revalution™
control software which features simple
intuitive menus and controls. Personalize
and store individual operators screen layouts
for smoother workflow. Customize filteringand image enhancements to suit yourapplications needs. Utilize the easy point& click controls to analyze defects fast andproduce automated inspection routines.

Optimal Clarity

Designed in-house by a team ofsemiconductor experts Onyx™ delivers stateof-the-art performance. Using world-classpixel specifications for low dose sensitivity& high saturation dose. Onyx™ has higher
frame rates and lower noise, giving you
faster clearer images.
Onyx HL Detector

Unique patented technology

QuadraNT® is the powerhouse of
Quadra Pro. Nordson’s latest generation
X-ray source gets another enhancementwith patent-pending, automatic Dual Modein the Quadra 7 Pro. The tube automaticallyswitches between two modes:• High resolution mode ideal for sensitive,
semiconductor samples• High flux mode essential for dense,
thicker samples
The benefit of Dual Mode is crystal clear
images to spot defects in more detail thanyou have seen before! The best-in-class
X-ray tube maintains its 10-fold brightness
and improved stability over conventional
tubes and bests them again with smaller
spot size at any voltage. The results you
can see for yourself.
SystemResolution6.5 MP

Footprint

1570 W x 1500 D x 1900 H mm

Frame Rate

30 fpsSystem Weight1950 kg (4300 lbs)

Image bit depth

16-bitPower Supply

Single phase 200-230 VAC,
50/60 Hz, 16A

Typical Power
Consumption

1kWAir Supply

5-8 bar clean, dry for
anti-vibration

Operation
Temperature

10-30°C

Humidity<85 % (non-condensing)Operation

Mouse point and click

Operating System

Windows 10 64-bit

X-ray Safety<1 μSv/hr Meets all
international standards

QuadraNTTM X-ray Tube
Type

Mk4 Dual Mode 20W

Feature Recognition<100nm (at less than 10W)

Max. Power

20 W

Voltage30-160kvInspection
Oblique Angle View

2 × 70° - No sample rotation required

Anti Vibration

ActiveInspection Area

510 x 445 mm (20 x 17.5”)

Geometric
Magnification

3000x

Total Magnification

68,000x

Display40” Ultra Wide Monitor (5120 x 2160)

www.nordson.com/TestInspect
DS-Q7PRO-090623

Options
X-Plane®

View sample slices without
additional CT hardware

μCT

Create high resolution 3D models
with the μCT sample stage

Barcode Reader

Track samples easily

Filtering Tray

Reduce sample X-ray dose

Wafer Holders

For handling 300 mm and200 mm wafers

For more information,
speak with your Nordson
representative or contact
your Nordson regional office
Nordson Test & Inspection
Europe, SEA, Africa
ti-sales-eu@nordson.com
Nordson Test & Inspection
Americasti-sales-us@nordson.com
Nordson Test & Inspection
Chinati-sales-cn@nordson.com
Nordson Test & InspectionJapanti-sales-jp@nordson.com
Nordson Test & Inspection
Singaporeti-sales-eu@nordson.com
Nordson Test & Inspection
Taiwanti-sales-tw@nordson.com
Nordson Test & Inspection
Korea
ti-sales-korea@nordson.com