CORDLES S B R OA DB A ND LE D CURI N G L I G H T

VALO™ GRANDEN

Curing Light

Product Description

With its broadband spectrum, the VALO Grand curing light is designed to polymerize all light cured products in the wavelength range of 385-515nm per ISO 10650. The VALO Grandcuring light uses the Ultradent VALO rechargeable batteries and battery charger. The curing light is designed to rest in a standard dental unit bracket or can be custom mounted using
the bracket included with the kit.

VALO Grand Product Components:
•
•
•
•
•
•

1 – VALO Grand curing light
4 – Ultradent VALO rechargeable batteries
1 – Ultradent VALO battery charger with medical grade 12VDC AC power adapter
1 – VALO Barrier Sleeve sample pack1 – VALO Grand light shield
1 – Curing light surface mounting bracket with double stick adhesive tape

Overview of Controls:

Photo

Buttons and Lights

LegendPower Button(s)(Top or Bottom)
Timing Lights
Low Battery - Indicator
Mode/Status Light
Timing/Mode Change
Button(Bottom)(Top)

For all products described, carefully read and understand all instructions and SDS information prior to use.

Indications for Use/Intended Purpose

The source of illumination for curing photo-activated dental restorative materials and adhesives.

Warnings and Precautions
Risk Group 2
CAUTION UV emitted from this product. Eye or skin irritation may result from exposure. Use appropriate shielding.
CAUTION Possibly hazardous optical radiation emitted from this product. Do not stare at operating lamp. May be harmful to the eyes.
• DO NOT look directly into the light output. Patient, clinician, and assistants should always wear amber colored UV eye protection when curing light is in use.
• To prevent the risk of electric shock, no modification of this equipment is allowed. Use only the included Ultradent VALO light power supply and plug adapters. If these components
are damaged, do not use and call Ultradent Customer Service to order a replacement.
• Portable RF communications equipment may degrade performance if used closer than 30 cm (12 in”).
• Use only authorized accessories, cables, and power supplies to prevent improper operation, increased electromagnetic emissions or decreased electromagnetic immunity (refer toElectromagnetic Emissions section).
• To avoid the risk of electrical fire associated with handling of batteries:
⸰ DO NOT mix rechargeable batteries with non-rechargeable batteries or other battery types.
⸰ DO NOT attempt to charge non-rechargeable batteries. Charging non-rechargeable batteries can result in injury or property damage.
⸰ DO NOT autoclave or spray batteries, as welbattery contacts, charger, or AC power adapter with liquid of any kind. If corrosion appears on the contacts of the battery charger, call
Ultradent Customer Service to order a replacement.
⸰ DO NOT charge batteries around flammable materials.
⸰ DO NOT keep charger in clinical operatory.
• To avoid the risk of injury, DO NOT use batteries that are corroded (rust), dented, emit an odor or fluids, have a torn or missing wrapping, or are otherwise damaged. Call Ultradent
Customer Service to order replacement batteries.
• Only use recommended batteries. Alternative batteries can cause malfunction.
• To prevent the risk of thermal irritation or injury, avoid back-to-back curing cycles and do not expose oral soft tissues at close proximity for more than 10 seconds in any mode. If
longer curing times are required, use multiple curing cycles with periods of rest between cycles, or use a dual-cure product to avoid heating soft tissue.
• Use caution when treating patients who suffer from adverse photobiological reactions or sensitivities, patients who are undergoing chemotherapy treatment, or patients being
treated with photosensitizing medication.
• This unit may be susceptible to strong magnetic or static electric fields, which could disrupt the programming. If you suspect this has occurred, unplug the unit momentarily and thenre-plug it into the outlet.
• DO NOT wipe down the curing light with caustic or abrasive cleaners, autoclave, or immerse in any kind of ultrasonic bath, disinfectant, cleaning solution, or liquid. Failure to follow
included processing instructions may render curing light inoperable.
• To avoid damaging the equipment, DO NOT insert fingers, instruments, or other objects into the battery compartment of the curing light.
3

•
•
•
•

To avoid damaging the equipment, DO NOT attempt to clean the gold contacts, or any part of the battery compartment. Call Ultradent Customer Service if there is a concern.
To prevent the risk of cross-contamination, barrier sleeves are single use.
To reduce the risk of corrosion, remove barrier sleeve after use.
To reduce the risk of under-cured resins, do not use curing light if lens is damaged.

Stepwise Instructions
Preparation

1) Charge batteries before using the curing light (See section Battery Maintenance).
2) Prior to each use, place a new barrier sleeve over the curing light, minimizing wrinkles over the lens for best results..
Note:• To help prevent cross-contamination and help keep dental composite material from adhering to the surface of the lens and wand body, an Ultradent approved barrier sleeve must be
used over the VALO curing light with each use. Barrier sleeves are intended for single-patient use.

Quick Mode Guide:
Standard Power
1000 mW/cm2

ModeHigh Power Plus
1600 mW/cm2

Xtra Power
3200 mW/cm2

Power Button
Mode/Timing LEDs

Time Buttons

VALO Cordless Light Shield:

The VALO Cordless Light Shield is oval-shaped, can be rotated for maximum use, and can be used with a transparent barrier sleeve.

UseTime Options

5s

10s

15s

20s

1s

2s

3s

4s

1) Each power mode is used for the curing of dental materials with photo initiators. See Quick Mode Guide for recommended curing times.
NOTE: The curing light is programmed to cycle from the Standard Power to the High-Power Plus to the Xtra Power mode in sequence. For example, to change from the Standard Powermode to the Xtra Power mode, it is necessary to cycle into the High-Power Plus mode and then to the Xtra Power mode.
2) The curing light stores the most recentely used timing inverval and mode, and it will default back to this whenever the modes are changed or if the batteries are removed.

To Change Time

Press and release Time Button quickly to cycle through time options.

To Change Modes

Press and hold Time Button for 2 seconds and release. Curing light will cycle to next Mode.

Operation
CURING MODE: Standard PowerMode

LegendSolid LEDs

TIMING INTERVALS: 5, 10, 15, and 20 seconds.
• The curing light defaults to this mode when it is INITIALLY powered on. The Mode/Status Light will be green and the four green Timing Lights are illuminated, indicating Standard
Power mode.
• To change timing intervals, quickly press the Time/Mode Button.
• Press the Power Button to cure. To stop curing prior to completion of a timing interval, press the Power Button again.
TIMING INTERVALS: 1, 2, 3, and 4 seconds.
• From Standard Power mode, press and hold the Time/Mode Change Button for 2 seconds and release. The Mode/Status Light will be orange, and the four green Timing Lights will
illuminate and flash, indicating High Power mode.
• To change timing intervals, quickly press the Time/Mode Button.
• Press either Power Button to cure. To stop curing prior to the completion of a timing interval, press either Power Button again.
• To return to Standard Power mode, press and hold the Time/Mode Change Button for 2 seconds and release, this will cycle to Xtra Power mode. Press and hold again for 2 seconds,
and release. The Mode/Status Light will be green and the four green Timing Lights are illuminated, indicating Standard mode.
TIMING INTERVAL: 3 seconds only (Note: The Xtra Power mode has a 2 second safety delay at the end of each curing cycle to limit heating during consecutive curing. At the end of thedelay, beeping indicates unit is ready for continued use).
• From Standard Power mode, press the Time/Mode Change Button for 2 seconds, release, press and hold again for 2 seconds, and release. The Mode/Status Light will be orange andflash, and three of the green Timing Lights will illuminate and flash, indicating Xtra Power mode.
• Press the Power Button to cure. To stop curing prior to the completion of a timing interval, press the Power Button again.
• To return to the Standard Power mode, press and hold the Time/Mode Button for 2 seconds and release. The Mode/Status Light will be green and the green Timing Lights are
illuminated, indicating Standard Power mode.

Sleep Mode: The curing light will go into Sleep Mode after 60 seconds of inactivity, as indicated by a slow flashing of the mode/status light. Picking up or touching the unit will wake-up
the curing light and automatically return it to the last setting used. To maximize battery life, leave the curing light undisturbed when not in use.

Blinking LEDs

Quick Curing Guide:

CURING MODE: High Power Plus Mode

CURING MODE: Xtra Power Mode

3s OnlyCuring Mode

Power* (mW)

Irradiance* (mW/cm2)

Total Exposure Time (Seconds)

Energy** (Joules)

Standard

970

900

5/10/15/20

4.8J/9.7J/14.5J/19.4J

High Power Plus

1,620

1,500

1/2/3/4

1.6J/3.2J/4.8J/6.5J

Xtra

2,260

2,100

3

6.8J

*Nominal output at a distance of 2 mm from the tip of the lens to the top surface of the composite.
**The values in the total energy per cycle (joules) section are rounded up to the nearest tenth.

Quick Warning Guide:
Power Level Warning

Temperature Warning

Service Warning

LED Warning

Replace batteries

Allow for cool down

Call Customer Service for Repair

Call Customer Service for Repair• No sound• Flashing, 2 seconds• Allows operation• Continuous 3 beeps• Single rapid flash• Prohibits operations• Low battery: slow flashing• Shut off: 3 beeps, flashing• Prohibits operation• 3 beeps• Slow flashing• Prohibits operation

Cleanup

1) Discard used barrier sleeves in standard waste after each patient.
2) See Processing Section.

Mounting Bracket Instructions

1) Bracket should be mounted to a flat, oil-free surface.
2) Clean surface with rubbing alcohol.
3) Peel backing off the bracket’s adhesive tape.
4) Position bracket so the curing light lifts upward when removed. Press firmly into place.

Maintenance

General Cleaning of Curing Light
After each use, moisten a gauze or soft cloth with an approved surface disinfectant and wipe the surface and lens. Unauthorized cleaners may cause damage to the curing light.
ACCEPTABLE CLEANERS:• 70% Isopropyl alcohol• 70% Ethanol

4General Cleaning of the Light Shield:
Clean the VALO Cordless light shield using any surface disinfectant. DO NOT autoclave.
RepairUser-Performed Maintenance
1) Use a barrier sleeve to prevent dental composite from adhering to the surface of the lens. If necessary, use a plastic or stainless steel dental instrument to carefully remove any
adhered composite. Do not use tools that will damage the lens.
52) Light meters differ greatly and are designed for specific light guide tips and lenses. Ultradent recommends routinely checking the output in Standard Power mode. NOTE: the true
numeric output will be skewed due the inaccuracy of common light meters and the custom LED pack in the curing light
Battery Maintenance
Charging and Changing Batteries
The curing light comes with 4 rechargeable lithium iron phosphate batteries. Ultradent recommends using VALO rechargeable batteries only.

How to change/insert batteries:
1) Remove back cap by twisting counterclockwise one quarter turn.
2) Remove batteries.
3) Insert fresh batteries positive (+) side first.
4) Reattach back cap by aligning and gently pushing while twisting clockwise. The cap will click when fully attached.
5) The unit is ready for use.
Low Batteries: The curing light signals the user that it is time to change the batteries when the Low Battery Indicator Light is flashing red. If the battery charge becomes too low, an
audible 3 beep warning sound will occur and the curing light will not allow further operation until batteries are recharged or new batteries are inserted. (See Quick Warning Guide)
Charging Time and Battery Life: Fully charged battery life in the curing light is dependent on the Mode/Time interval, battery type, amount of use, and LED efficiency.
• Recommended Recharge Interval: Recharge as necessary or, when low battery indicator comes on. Life expectancy depends on use.
• Extra Batteries: The curing light comes with 4 rechargeable batteries.
Manufacturer Repair
1) Repairs are only to be performed by authorized service personnel. Ultradent is to provide service personnel with documentation to perform repairs.
2) When sending units in for repair, service, or calibrations always remove the batteries from the curing light and charger. Wrap batteries, charger, adapter, and the curing light
separately in the return box.
3) Ship batteries in accordance with local regulations.
Warranty
Ultradent Products, Inc. (“Ultradent”) warrants that this product shall, for a period of 5 years from the date of purchase, when operated according to the operating instructions included
with the product, (i) conform in all material respects to the specifications set forth in Ultradent’s documentation accompanying the product; and (ii) be free from defects in material andworkmanship.
This limited warranty is nontransferable and applies solely to the original purchaser and does not extend to subsequent owners of the product. This limited warranty does not coverany other accessory components such as, but not limited to, batteries, chargers, adapters, or adaptive lenses. This limited warranty is void if the product fails or is damaged due tonegligence, abuse, misuse, accident, modification, tampering, alteration, or failure to follow the applicable instructions for use. For purposes of example only, a product that is dropped
and damaged is not covered under this warranty. To qualify under this limited warranty, proof of purchase (e.g., sales receipt or similar documentation) must be submitted to Ultradentalong with the defective product.
A defective product meeting the warranty conditions set forth herein will, at Ultradent’s sole discretion, either be repaired or replaced. In no event shall Ultradent’s liability for the
product exceed the purchase price paid by the purchaser. Under no circumstances shall Ultradent be liable for any indirect, incidental, foreseen, unforeseen, special, or consequential
damages arising out of or in connection with the use of this product.

Processing

VALO Batteries

Technical information/Data

Effective Composite Curing Wavelength Bands:
EFFECTIVE COMPOSITE CURING WAVELENGTH BANDS(Utilizable Wavelength Range: 385-515 nm)
Effective lower
wavelength band
395-415 nm

Attribute

Information/Specification

Lens

Diameter 11.7 mm

Wavelength
range

When disposing of electronic waste (i.e. curing lights, chargers, batteries and power supplies), follow local waste and recycling guidelines.

6• Utilizable wavelength range: 385 – 515nm• Peak wavelengths: 395 – 415nm and 440 – 480nm

Measurement Instrument
Light Intensity
Table

Technical Considerations
Accessories

VALO Cordless Light Shield

High Power

Nominal Radiant Exitance Comparison Chart

If storing the curing light for periods longer than 2 weeks, or packing it for travel, always remove the batteries. If batteries are left in the unit for long periods of time without recharging
they may become nonfunctional or un-chargeable. Do not store batteries in temperatures over 40°C (104°F) or in direct sunlight.

CE Information

Xtra

WAVELENGTH (nm)This product is intended to be used with a barrier sleeve. See above for maintenance and general cleaning.

Item

Effective upper
wavelength band
440-480 nm

Standard

Storage and Disposal

Curing light Storage and Transport:• Temperature: +10°C to +40°C (+50°F to +104°F)• Relative Humidity: 10% to 95%• Ambient Pressure: 500 hPa to 1060 hPa

CE Information

VALO Charger

SPECTRAL RADIANT POWER (mW/nm)

How to charge batteries:
1) Place charger and batteries in an area where they will not come into contact with liquids or disinfectant spray.
2) Insert batteries into charger with positive (+) end pointed towards the indicator lights on charger.
3) Green Lights indicate batteries are ready for use.
4) Batteries will take 1-3 hours to charge. Leave batteries in charger until ready for use.
NOTE: If the red light on the charger does not turn green when the batteries have been charging longer than three hours, the battery may be degraded (corrupted) and cannot becharged. Try a new battery or call Ultradent Customer Service to order a new set of rechargeable batteries.

Item‡ Gigahertz spectrum analyzer
Exitance

Total Power

Aperture of Meter

15 mm

15 mm

Standard Power (±10%)900 mW/cm²

970 mW

High Power Plus (±10%)1500 mW/cm²

1615 mW

Xtra Power (±10%)2100 mW/cm²

2260 mW

Radiant Exitance will vary based on instrument
capability, measurement method, and lightplacement.
‡ Radiant Exitance conforms to ISO 10650 whenmeasured with a Gigahertz-Optik spectrum analyzer.

VALO Grand
Curing Light

Weight:• With batteries: 6 oz. (190 g)
Ratings: IEC 60601-1 (Safety), IEC 60601-1-2 (EMC)• Without batteries: 5 oz. (150 g)
Dimension: (8 x 1.28 x 1.06) inches, (203 x 32.5 x 27) mm

Charger Power
SupplyOutput - 12VDC at 500mA
Input - 100VAC to 240VAC
Ultradent P/N 5930 VALO Charger Power Supply
with international plug inserts

Ratings: IEC 60601-1 (Safety)
Cord Length - 6 ft (1.8 m)
VALO Charger Power Supply is a Medical Grade Class II power supply andprovides isolation from MAINS power
7

Attribute

Information/Specification

Guidance and Manufacture’s Declaration for Electromagnetic Emissions
RF emissions
CISPR 11

Group 1

The curing light uses batteries and is not affected by EMI, RF, or surge suppression.

VALO Charger

VALO 3.6VDC Lithium Iron Phosphate smart battery charger:• Automatic shut off when fully charged• Auto-detection of defective batteries• Protections: Overcharge, Short-circuit, reverse polarity
⸰ Red LED – Charging
⸰ Green LED – Empty or Fully Charged
⸰ LED off – short circuit• Charging time: 1 – 3 hours
Rating: CE, WEEE

RF emission
CISPR 11Class BThe curing light uses electrical and electromagnetic energy only for their internal functions.
However, any RF emissions are very low and are not likely to cause interference in nearby
electronic equipment.

Harmonic emissions
IEC 61000-3-2

N/A

Harmonic emission and voltage fluctuation testing are not applicable to the curing light because
it is battery powered.

Voltage fluctuations/flicker
emissions IEC 61000-3-3

N/A

The curing light is suitable for use in all establishments, including domestic establishments andthose directly connected to the public low-voltage power supply network that supplies buildingsfor domestic use.

VALO Batteries

Rechargeable: Safe chemistry Lithium Iron Phosphate (LiFePO4) RCR123A• Working Voltage: 3.2VDC, 400mAh (Watt hour needed)
Ratings: UL 62133-2, CE, RoHS, WEEE
Cells are rated for >2000 charge/discharge cycles before hitting 70% of its capacity. It is advisable to replace batteries every 3-5 years,
but life may vary depending on use. Replace as necessary.

Operating,Charging
Conditions

Temperature: +10°C to +32°C (+50°F to +90°F)
Relative Humidity: 10% to 95%
Ambient Pressure: 700 hPa to 1060 hPa

The curing light is intended for use in the electromagnetic environment specified below. The customer or user should ensure that it is used in such an environment.
IMMUNITY test

IEC 60601 test level

Compliance level

Electromagnetic environment guidance

Duty Cycle

The curing light is designed for short-term operation. At maximum ambient temperature (32°C) 1 minute ON back-to-back cycling, 30
minutes OFF (cooling-off period).

Electrostatic discharge (ESD)± 8 kV contact± 15 kV air± 8 kV contact± 15 kV air

Physical environment should be restricted to thefollowing:
1) IP Code: IP20
2) Do not immerse in liquid.
3) Do not use around flammable gas. Unit is NonAPG and Non-AP.
4) Storage humidity range: 10%-95%
5) Storage temperatures range: 10° C-40° C

Electrical fast transient/burst
IEC 61000-4-4± 2 kV for power supply lines± 1 kV for input/output lines± 2 kV for power supply lines
Note 1: The curing light has noI/0 ports

The curing light is battery powered and is not capableof connection to AC MAINS power.

Surge
IEC 61000-4-5± 1 kV line to line
± 2 kV line to earth± 1 kV line to line
± 2 kV line to earth

Voltage, dips, shorts,
interruptions, and variations on
the power supply input lines<5% U(>95% dip in U for 0.5 cycle)<5% U(>95% dip in U for 0.5 cycle)

40% U(60% dip in U for 5 cycles)

40% U(60% dip in U for 5 cycles)

70% U(30% dip in U for 25 cycles)

70% U(30% dip in U for 25 cycles)<5% U(>95% dip in U for 5 s)

<5% U(>95% dip in U for 5 s)Note 2: Self recovers

30 A/m30 A/mGuidance and Manufacture’s Declaration for Electromagnetic Immunity

IEC 61000-4-2

Trouble Shooting
If the solutions suggested below do not rectify the problem, please call Ultradent at 800.552.5512. Outside the United States, call your Ultradent distributor or dental
dealer.
ProblemPossible Solutions

Light will not turn on1) Press the Time/Mode or Power Button to wake from Power Save Mode.
2) Check the red Low Battery Indicator for battery charge status.
3) Check that fresh batteries are correctly inserted into the unit.
4) If red and yellow Warning LEDs are flashing this means the curing light has reached its internal temperature safety limit.
Allow the curing light to cool down for 10 minutes or use a cool moist towel to cool the unit down quickly.
5) If red Warning LED flashes and beeps continuously, call Ultradent Customer Service for repair.

Light does not stay on for desired time1) Check Mode and Timing lights for correct time input.
2) Check the Low Battery Indicator for battery charge status.
3) Check that fresh batteries are properly inserted into the unit.

Light is not curing resins properly

1) Check lens for residual cured resins/composites.
2) Using proper amber UV eye protection, verify the LED lights are working.
3) Check power level with light meter. If using a light meter, Ultradent recommends checking the curing light in Standard
Power mode.
NOTE: The true numeric output will be skewed due to the inaccuracy of common light meters and the custom LED pack the
curing light uses. Light meters differ greatly and are designed for specific light guide tips and lenses.
4) Check expiration date on curing resin.
5) Ensure proper technique is being followed according to manufacturer recommendations.

Batteries will not charge

1) Make sure batteries are inserted into the charger in the correct orientation and allow batteries to charge for 1-3 hours.
2) If red lights on the charger do not change to green, call Ultradent Customer Service to order replacement batteries and/or charger.
3) If neither green nor red lights on the charger are visible, call Ultradent Customer Service to order or replace charger
and/or AC adapter.

Charger does not charge batteries

1) Make sure charger is plugged in and AC adapter is plugged into a working power outlet.
2) If green or red lights on the charger are not visible, call Ultradent Customer Service for new charger and/or AC adapter.

Cannot change mode or time intervals

1) Hold both Time/Mode and Power buttons down until a series of beeps indicates the curing light is unlocked.

IEC 61000-4-11

Power frequency(50/60 Hz) magnetic field
IEC 61000-4-8

Because the curing light is battery powered, it is not
subject to electric transients, surges, voltage dips,
shorts, interruptions, or variations on AC MAINS
power.
The accessory battery charger may be subject to theabove but it is separate from and not critical to theoperation of the VALO Cordless curing light.
If battery voltage of the curing light drops to 4VDC theunit will not allow operation. The VALO Cordless curing
light will turn off. When new batteries are put in andthe proper power levels are restored, the curing lightwill restart and return to the same state before power
loss. The curing light will self-recover in the eventof power loss.
Power frequency magnetic fields should be atlevels characteristic of a typical location in a typical,
residential, home health care, commercial, hospital, or
military environment.

Note 1: The curing light is not equipped with any ports or any accessible I/O lines.
Guidance and Manufacture’s Declaration for Electromagnetic Immunity for non-life support systems
The curing light is intended for use in the electromagnetic environment specified below. The customer or user should ensure that it is used in such an environment.
IMMUNITY test

IEC 60601 test level

Compliance level

Electromagnetic environment guidance

Miscellaneous Information
Guidance and Manufacture’s Declaration for Electromagnetic Emissions
The curing light is intended for use in the electromagnetic environment specified below. The customer or user should ensure that it is used in such an environment.
Emissions Test
8

ComplianceElectromagnetic environment - guidance
9

Report any serious incident to the manufacturer and the competent authority.
www.ultradent.com / 1-800-552-5512 / 801-572-4200

Guidance and Manufacture’s Declaration for Electromagnetic Immunity for non-life support systems

Conduction RF

3 Vrms3 VrmsIEC 61000-4-6

150 kHz to 80 MHz

150 kHz to 80 MHz

Radiated RF

3 V/m

3 V/m

Portable and mobile RF communications equipment should be used no closer to any
part of the VALO GRAND, including cables, than the recommended separation distancecalculated from the equation applicable to the frequency of the transmitter.
Recommended separation distance:

80 MHz to 800 MHz

800 MHz to 2.5 GHz
P is the maximum output power rating of the transmitter in watts (W) according to thetransmitter manufacturer and d is the recommended separation distance in meters (m).
Field strengths from fixed RF transmitters, as determined by an electromagnetic site
surveya, should be less than the compliance level in each frequency rangeb..
IEC 61000-4-3

80 MHz to 2.5 GHz

80 MHz to 2.5 GHzInterference may occur in the vicinity of equipment marked with the following symbol:

NOTE 1 At 80 MHz and 800 MHz, the higher frequency range applies.
NOTE 2 These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorption and reflection from structures, objects and people.
Field strengths from fixed transmitters, such as base stations for radio (cellular/cordless) telephones and land mobile radios, amateur radio, AM and FM
radio broadcast and TV broadcast cannot be predicted theoretically with accuracy. To assess the electromagnetic environment due to fixed RF transmitters,
an electromagnetic site survey should be considered. If the measured field strength in the location in which the curing light is used exceeds the applicable RFcompliance level above, the VALO Cordless curing light should be observed to verify normal operation. If abnormal performance is observed, additional measures
may be necessary, such as re-orienting or relocating the curing light.
bOver the frequency range 150 kHz to 80 MHz, field strengths should be less than 3 V/m.
a

Guidance and Manufacture’s Declaration for recommended separation distances between portable and mobile RF communications equipment and the VALO
Cordless curing light
The curing light is intended for use in an electromagnetic environment in which radiated RF disturbances are controlled. The user of the curing light can help prevent
electromagnetic interference by maintaining a minimum distance between portable and mobile RF communications equipment (transmitters) and the curing light asrecommended below, according to the maximum output power of the communications equipment.
Separation distance according to frequency of transmitter(meters)
Rated maximum output powerof transmitter(P in Watts)

150 kHz – 80 MHz

80 MHz – 800 MHz

800 MHz – 2.5 GHz

0.01

0.12 meters0.035 meters0.07 meters

0.1

0.37 meters0.11 meters0.22 meters1

1.7 meters0.35 meters

0.7 meters

10

3.7 meters1.11 meters2.22 meters100

11.7 meters

3.5 meters

7.0 meters

The curing light has been tested according to IEC 60601-1-2:2014 and passed under radiated field strengths of 10 V/m between 80-MHz to 2.5 GHz. The value of
3Vrms corresponds to V1 and the value 10 V/m corresponds to E1 in the formulas above.
For transmitters rated at a maximum output power not listed above, the recommended separation distance d in meters (m) can be estimated using the equation
applicable to the frequency of the transmitter, where P is the maximum output power rating of the transmitter in watts (W) according to the transmitter
manufacturer.
NOTE 1 At 80 MHz and 800 MHz, the separation distance for the higher frequency range applies.
NOTE 2 These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorption and reflection from structures, objects and people.
10

11

VALO™ GRANDDE

POLYMERISATIONSLEUCHTE
Product Description

Mit seinem breitbandigen Spektrum ist das VALO Grand Härtungslicht so konzipiert, dass es alle lichtgehärteten Produkte im Wellenlängenbereich von 385-515 nm gemäß ISO 10650
polymerisiert. Das VALO Grand-Härtungslicht verwendet die Ultradent VALO Akkus und das Batterieladegerät. Das Härtungslicht ist so konzipiert, dass sie in einer Standardhalterung fürzahnärztliche Einheiten ruht, oder kann mithilfe der im Lieferumfang enthaltenen Kit individuell montiert werden.

VALO Grand Produktkomponenten:
•
•
•
•
•
•

1 – VALO Grand-Härtungslicht
4 – Ultradent VALO wiederaufladbare Batterien
1 – Ultradent VALO Batterieladegerät mit medizinischem 12VDC AC-Netzanschluss
1 - VALO Barrierehülle Musterpackung
1 – VALO Grand-Lichtschild
1 – Halter für die Oberflächenmontage des Lichthärtegeräts mit DoppelklebebandÜbersicht über die Bedienelemente:

Foto

VALO Cordless Lichtschutzschirm:
Tasten und Lampen

LegendeTiming-Kontrollampen
Anzeige für niedrige Batterieladung
Modus-/Status-Kontrolllampe
Taste Timing/Modus ändern(Oben)

Alle Anleitungen und SDB-Informationen für alle beschriebenen Produkte vor Verwendung dieser Produkte sorgfältig lesen und verstehen.

Gebrauchshinweise / Verwendungszweck

Die Beleuchtungsquelle für die Aushärtung von fotoaktivierten Zahnrestaurationsmaterialien und Klebstoffen.

Warnhinweise und Vorsichtsmaßnahmen
Risikogruppe 2
CVORSICHT! Dieses Produkt emittiert UV-Strahlung. Kontakt kann zu Augen- oder Hautreizungen führen. Geeignete Abschirmung verwenden.
VORSICHT! Möglicherweise wird von diesem Produkt gefährliche optische Strahlung emittiert. Richten Sie Ihren Blick nicht auf die Betriebsleuchte. Dies kanngesundheitsschädlich für die Augen sein.
• Schauen Sie NICHT direkt in die Lichtquelle. Patienten, Kliniker und Assistenten sollten immer bernsteinfarbenen UV-Augenschutz tragen, wenn Licht zum Aushärten verwendet wird.
• Um die Gefahr eines Stromschlags zu vermeiden, sind keine Änderungen an diesem Gerät zulässig. Verwenden Sie nur die mitgelieferten Ultradent VALO Netzteile undKabelsteckeradapter. Wenn diese Komponenten beschädigt sind, verwenden Sie es nicht und wenden Sie sich an den Ultradent-Kundendienst, um einen Ersatz zu bestellen.
• Portable HF-Kommunikationsgeräte können die Leistung beeinträchtigen, wenn sie näher als 30 cm (12 Zoll) verwendet werden.
• Verwenden Sie nur zugelassene Zubehör, Kabel und Netzteile, um unsachgemäßen Betrieb, erhöhte elektromagnetische Emissionen oder verringerte elektromagnetische
Störfestigkeit zu verhindern (Siehe hierzu Abschnitt Elektromagnetische Emissionen).
• Um die Gefahr eines elektrischen Brandes im Zusammenhang mit dem Umgang mit Batterien zu vermeiden:⸰ Mischen Sie KEINE wiederaufladbaren Batterien mit nicht wiederaufladbaren Batterien oder anderen Batterietypen.
⸰ Versuchen Sie NICHT, nicht wiederaufladbare Batterien aufzuladen. Das Aufladen von nicht wiederaufladbaren Batterien kann zu Verletzungen oder Sachschäden führen.
⸰ Batterien, Batteriekontakte, das Ladegerät oder der Netzadapter dürfen NICHT autoklaviert oder mit Flüssigkeiten besprüht werden. Wenn an den Kontakten desBatterieladegeräts Korrosion auftritt, wenden Sie sich an den Ultradent-Kundendienst, um einen Ersatz zu bestellen.
⸰ Laden Sie Batterien NICHT in der Nähe von brennbaren Materialien auf.
⸰ Lassen Sie das Ladegerät NICHT im klinischen Betrieb.
• Um Verletzungsgefahr zu vermeiden, verwenden Sie KEINE Batterien, die korrodiert (Rost), verbeult sind, einen Geruch oder Flüssigkeiten aufweisen, eine zerrissene oder fehlendeVerpackung haben oder anderweitig beschädigt sind. Rufen Sie den Ultradent-Kundendienst an, um Ersatzbatterien zu bestellen.
• Verwenden Sie nur empfohlene Batterien. Alternative Batterien können zu Fehlfunktionen führen.
• Um das Risiko von thermischen Irritationen oder Verletzungen zu vermeiden, vermeiden Sie aufeinanderfolgende Härtungszyklen und setzen Sie orales Weichgewebe nicht länger
als 10 Sekunden in irgendeinem Modus in unmittelbarer Nähe aus. Wenn längere Aushärtezeiten erforderlich sind, verwenden Sie mehrere Aushärtezyklen mit Pausen zwischen den
Zyklen oder verwenden Sie ein Dual-Cure-Produkt, um ein Erhitzen von Weichgewebe zu vermeiden.
• Seien Sie vorsichtig bei der Behandlung von Patienten, die an unerwünschten photobiologischen Reaktionen oder Empfindlichkeiten leiden, Patienten, die sich einer
Chemotherapiebehandlung unterziehen, oder Patienten, die mit photosensibilisierenden Medikamenten behandelt werden.
• Dieses Gerät kann anfällig für starke magnetische oder statische elektrische Felder sein, die die Programmierung stören könnten. Wenn Sie vermuten, dass dies der Fall ist, ziehen
Sie das Gerät kurz aus der Steckdose und stecken Sie es dann wieder in die Steckdose ein.
• Wischen Sie das Härtungslicht NICHT mit ätzenden oder scheuernden Reinigungsmitteln ab, autoklavieren Sie es nicht und tauchen Sie es nicht in ein Ultraschallbad,
12

Schrittweise Anleitung
Vorbereitung

1) Laden Sie die Batterien auf, bevor Sie das Härtungslicht verwenden (siehe Abschnitt Batteriewartung).
2) Legen Sie vor jedem Gebrauch eine neue Schutzhülle über das Aushärtelicht und minimieren Sie Knitterfalten über der Linse, um beste Ergebnisse zu erzielen.
Hinweis:• Um Kreuzkontaminationen zu vermeiden und zu verhindern, dass dentales Kompositmaterial an der Oberfläche der Linse und des Stabkörpers haftet, muss bei jeder Verwendungeine von Ultradent zugelassene Barrierehülse über dem VALO Härtungslicht verwendet werden. Barrierehülsen sind für den Einzelpatientengebrauch vorgesehen.

Einschalttasten(Oben oder unten)(Unten)Desinfektionsmittel, einer Reinigungslösung oder in eine Flüssigkeit. Die Nichtbeachtung der beigefügten Verarbeitungsanweisungen kann dazu führen, dass das Härtungslicht
funktionsunfähig wird.
• Um eine Beschädigung des Geräts zu vermeiden, führen Sie KEINE Finger, Instrumente oder andere Gegenstände in das Batterie der Aushärtelampe ein.
• Versuchen Sie NICHT, die Gold-Kontakte oder Teile des Batteries zu reinigen, um eine Beschädigung am Gerät zu vermeiden. Wenden Sie sich an den Ultradent-Kundendienst, wenn
Sie Bedenken haben sollten.
• Um das Risiko einer Kreuzkontamination zu vermeiden, dürfen die Barrierehüllen nur für einen Patienten verwendet werden.
• Um das Korrosionsrisiko zu verringern, entfernen Sie die Barrierehülse nach dem Gebrauch.
• Um das Risiko von unterhärtenden Harzen zu verringern, verwenden Sie kein Härtungslicht, wenn die Linse beschädigt ist.

Der VALO Cordless Lichtschutzschirm hat eine ovale Form, ist drehbar für optimale Verwendung, und kann mit einer transparenten Barriere-Schutzhülle verwendet werden.

Gebrauch

1) Jeder Lichtstärkemodus wird zum Aushärten von Dentalwerkstoffen mit Photoinitiatoren verwendet. Die empfohlenen Aushärtezeiten finden Sie in der Modus-Kurzanleitung.
HINWEIS: Die Polymerisationsleuchte ist so programmiert, dass sie der Reihe nach vom Modus Standard-Lichtstärke über den Modus Hohe Lichtstärke Plus in den Modus MaximalLichtstärke wechselt. Um beispielsweise vom Modus Standard-Lichtstärke in den Modus Maximal-Lichtstärke zu wechseln, ist zuerst ein Wechseln in den Modus Hohe Lichtstärke Plusund dann in den Modus Maximal-Lichtstärke erforderlich.
2) Die Polymerisationsleuchte speichert das zuletzt verwendete Zeitintervall und den zuletzt verwendeten Modus und wird einen Rücksprung auf diese Werte durchführen, wenn dieseModi gewechselt werden oder wenn die Akkus entfernt werden.

BedienungHÄRTUNGSMODUS: Modus Standard-Lichtstärke

ZEITINTERVALLE: 5, 10, 15, und20 Sekunden.
• Die Polymerisationsleuchte springt in diesen voreingestellten Modus, wenn es ERSTMALS eingeschaltet ist. Die Modus-/Status-Kontrollleuchte leuchtet grün und die vier grünen
Timing-Kontrollleuchten leuchten, was den Modus Standard-Lichtstärke anzeigt.
• Zum Ändern der Zeitintervalle schnell die Zeit-/Modus-Taste drücken.
• Zum Starten des Aushärtungsvorgangs auf die Einschalttaste drücken. Zum Beenden des Aushärtungsvorgangs vor Ablauf des Zeitintervalls die Einschalttaste erneut drücken.

HÄRTUNGSMODUS: Modus Hohe Lichtstärke PlusZEITINTERVALLE: 1, 2, 3, und 4 Sekunden.
• Im Modus Standard-Lichtstärke die Taste Zeit/Modus ändern 2 Sekunden lang drücken und dann loslassen. Die Modus-/Status-Kontrollleuchte leuchtet orange und die vier grünen
Timing-Kontrollleuchten leuchten auf und blinken, was den Modus Hohe Lichtstärke anzeigt.
• Zum Ändern der Zeitintervalle schnell die Zeit-/Modus-Taste drücken.
• Drücken Sie irgendeine Einschalttaste, um mit der Aushärtung zu beginnen. Um die Aushärtung vor Ablauf eines Zeitintervalls zu beenden, drücken Sie erneut die Einschalttaste.
• Taste Zeit/Modus ändern 2 Sekunden lang drücken und dann loslassen, um zum Modus Standard-Lichtstärke zurückzukehren. Dies wechselt in den Modus Maximal-Lichtstärke.
Taste erneut 2 Sekunden lang drücken und dann loslassen. Die Modus-/Status-Kontrollleuchte leuchtet grün und die vier grünen Timing-Kontrollleuchten sind beleuchtet, was denStandard-Modus anzeigt.

HÄRTUNGSMODUS: Modus Maximal-Lichtstärke

ZEITINTERVALL: Nur 3 Sekunden (Hinweis: Der Modus Maximal-Lichtstärke bietet am Ende jedes Aushärtungszyklus eine Sicherheitsverzögerung von 2 Sekunden, um bei
aufeinanderfolgenden Aushärtevorgängen ein Erhitzen zu begrenzen. Am Ende der Verzögerung signalisiert ein Signalton, dass das Gerät für den weiteren Gebrauch bereit ist).
• Im Modus Standard-Lichtstärke die Taste Zeit/Modus ändern 2 Sekunden lang drücken, loslassen, dann nochmals 2 Sekunden lang drücken und loslassen. Die Modus-/StatusKontrollleuchte leuchtet orange und blinkt, und drei der grünen Timing-Kontrollleuchten leuchten auf und blinken, was den Modus Maximal-Lichtstärke anzeigt.
• Zum Starten des Aushärtungsvorgangs auf die Einschalttaste drücken. Die Einschalttaste nochmal drücken, um die Aushärtung vor Ablauf des Zeitintervalls zu beenden.
• Die Zeit-/Modus-Taste 2 Sekunden lang drücken und dann loslassen, um zum Modus Standard-Lichtstärke zurückzukehren. Die Modus-/Status-Kontrollleuchte leuchtet grün und die
grünen Timing-Kontrollleuchten leuchten, was den Modus Standard-Lichtstärke anzeigt.

Energiesparmodus: Die Polymerisationsleuchte wechselt nach 60 Sekunden Inaktivität in den Energiesparmodus, was durch ein langsames Blinken der Modus-/Status-Kontrollleuchteangezeigt wird. Durch Aufnehmen oder Berührung des Geräts wird die Polymerisationsleuchte aktiviert und automatisch auf die zuletzt verwendete Einstellung zurückgesetzt. Die
Polymerisationsleuchte ungestört lassen, wenn es nicht verwendet wird, um die Akkulaufzeit zu maximieren.

Säuberung1) Gebrauchte Barriere-Schutzhüllen nach jedem Patienten als Restmüll entsorgen.
2) Siehe Abschnitt Verarbeitung.

Anleitung für die Montagehalterung

1) Die Halterung sollte auf einer flachen, ölfreien Oberfläche montiert werden.
2) Oberfläche mit Reinigungsalkohol reinigen.
3) Klebeband an der Rückseite der Halterung abziehen.
4) Halterung so positionieren, dass beim Entfernen die Polymerisationsleuchte nach oben abgehoben wird. Die Halterung an ihrem Platz fest anpressen.

Modus-Kurzanleitung:13

Standard-Lichtstärke
1000 mW/cm2

Modus

Modus Hohe Lichtstärke
1600 mW/cm2

Maximal-Lichtstärke
3200 mW/cm2

Einschalttaste
Modus-/Timing-LEDsUhrzeit-Tasten

Uhrzeit-Optionen

5s

10s

15s

20s

1s

2s

3s

4s

Nur 3sDie Uhrzeit ändern

Uhrzeit-Taste drücken und loslassen, um die Zeit-Optionen schnell zu durchlaufen.

Das Ändern von Modi

Uhrzeit-Taste 2 Sekunden lang drücken und dann loslassen. Die Polymerisationsleuchte wechselt in den nächsten Modus.

LegendeFeste LEDs

Blinkende LEDs

Kurzanleitung zur Aushärtung:
Härtungsmodus

Power* (mW)Bestrahlungsstärke* (mW/cm2)

Gesamtbelichtungszeit (Sekunden)

Energie** (Joule)Standard

970

900

5/10/15/20

4.8J/9.7J/14.5J/19.4J

High-Power Plus

1,620

1,500

1/2/3/4

1.6J/3.2J/4.8J/6.5J

Xtra

2,260

2,100

3

6.8J

*Nennleistung in einem Abstand von 2 mm von der Linsenspitze zur Oberseite des Verbundmaterials.
**Die Werte im Abschnitt Gesamtenergie pro Zyklus (Joule) sind auf das nächste Zehntel aufgerundet.

Kurzanleitung zu Warnhinweisen:
Ladezustandswarnung
Akkus ersetzen• Niedriger Batterie-Ladestand:langsam blinkend• Abschalten: 3 Signaltöne, blinkend• Verbietet den Betrieb

Temperaturwarnung

Service-Warnhinweis

LED-Warnung

Abkühlen lassen

Wenden Sie sich wegen einer
Reparatur an den Kundendienst

Wenden Sie sich wegen einer
Reparatur an den Kundendienst• 3 Signaltöne• Langsames Blinken• Verbietet den Betrieb

•
•
•
•

Kein Signalton
Blinken im
2 Sekundentakt
Gestattet den Betrieb• Durchgehend 3 Signaltöne• Einzelnes schnelles Blinken• Verbietet den Betrieb

Edelstahl, um angeklebte Verbundstoffe vorsichtig zu entfernen. Verwenden Sie keine Tools, die das Objektiv beschädigen können.
2) Lichtmessgeräte unterscheiden sich stark und sind für bestimmte Lichtleiterspitzen und -linsen konzipiert. Ultradent empfiehlt, die Leistung routinemäßig im Standard-Power-Moduszu überprüfen. HINWEIS: Die tatsächliche numerische Leistung wird aufgrund der Ungenauigkeit üblichen Lichtmessgeräte und des speziellen LED-Pakets im Lichthärtegerät verzerrt
sein.
Batteriewartung
Laden und Wechseln von Batterien
Das Härtungslichts wird mit 4 wiederaufladbaren Lithium-Eisen-Phosphat-Batterien geliefert. Ultradent empfiehlt, nur wiederaufladbare VALO-Batterien zu verwenden.
So laden Sie Batterien auf:
1) Platzieren Sie das Ladegerät und die Batterien an einen Ort, an dem sie nicht mit Flüssigkeiten oder Desinfektionssprays in Berührung kommen.
2) Legen Sie die Batterien in das Ladegerät ein, so dass der Pluspol (+) in Richtung der Anzeigeleuchten am Ladegerät zeigt.
3) Grüne Lichter zeigen an, dass die Batterien betriebsbereit sind.
4) Das Aufladen der Batterien dauert 1-3 Stunden. Lassen Sie die Batterien im Ladegerät, bis sie einsatzbereit sind.
HINWEIS: Wenn das rote Licht am Ladegerät nicht grün leuchtet u. die Batterien länger als drei Stunden geladen wurden, ist die Batterie möglicherweise beschädigt und kann nicht
geladen werden. Versuchen Sie es mit einer neuen Batterie oder rufen Sie den Ultradent-Kundendienst an, um einen neuen Set wiederaufladbarer Batterien zu bestellen.
So wechseln/platzieren Sie die Batterien:
1) Entfernen Sie die hintere Kappe, indem Sie sie eine Viertelumdrehung gegen den Uhrzeigersinn drehen.
2) Entfernen Sie die Batterien.
3) Legen Sie neue Batterien mit der positiven (+) Seite zuerst ein.
4) Bringen Sie die hintere Kappe wieder an, indem Sie sie ausrichten und vorsichtig drücken, während Sie sie im Uhrzeigersinn drehen. Die Kappe klickt, wenn sie vollständig sitzt.
5) Das Gerät ist einsatzbereit.
Niedriger Batteriestand: Das Härtungslicht signalisiert dem Benutzer, dass es an der Zeit ist, die Batterien zu wechseln, wenn die Kontrollleuchte für niedrigen Batteriestand rot
blinkt. Wenn die Batterieladung zu niedrig wird, ertönt ein akustisches Warnsignal mit 3 Pieptönen und das Lichthärtungsgerät kann nicht weiter betrieben werden, bis die Batterienaufgeladen oder neue Batterien eingelegt sind. (Siehe Schnell-Warnhinweis)
Nur VALO Kabellos/VALO Grand (NUR UD)
Ladezeit und Batterielebensdauer: Die Lebensdauer einer voll aufgeladenen Batterie im Härtungslicht ist vom Modus/Zeitintervall, dem Batterietyp, der Nutzungsdauer und der LEDEffizienz abhängig.
• Empfohlenes Aufladeintervall: Nach Bedarf aufladen oder wenn die Anzeige für niedrigen Batteriestand leuchtet. Lebensdauer hängt von der Nutzung ab.
• Zusätzliche Batterien: Das Härtungslicht wird mit 4 wiederaufladbaren Batterien geliefert.
VALO Ortho kabellos NUR (NUR UD)
Ladezeit und Batterielebensdauer: Die Lebensdauer einer voll aufgeladenen Batterie im Härtungslicht ist vom Modus/Zeitintervall, dem Batterietyp, der Nutzungsdauer und der LEDEffizienz abhängig. Im Allgemeinen sollten wiederaufladbare Batterien etwa 400 Härtungszyklen überstehen.
• Empfohlenes Aufladeintervall: Wenn die Anzeige für einen niedrigen Batteriestand aufleuchtet oder etwa alle 1 - 2 Wochen, je nach Nutzung.
• Zusätzliche Batterien: Das Härtungslicht wird mit 4 wiederaufladbaren Batterien geliefert.
• Batterielebensdauer: Wiederaufladbare Lithium-Eisen-Phosphat-Batterien können ca. 1000- bis 2000-Mal aufgeladen werden. Bei normalem Gebrauch und richtiger Pflege sollten
die Batterien bis zu fünf Jahre halten, aber bei Bedarf müssen sie ausgetauscht werden.
Reparatur durch Hersteller
1) Reparaturen dürfen nur von autorisiertem Servicepersonal durchgeführt werden. Ultradent stellt dem Servicepersonal Unterlagen zur Durchführung von Reparaturen zur Verfügung.
2) Wenn Sie Geräte zur Reparatur, Wartung oder Kalibrierung zuschicken, entfernen Sie immer die Batterien aus dem Härtungslichtgerät und dem Ladegerät. Wickeln Sie die Batterien,das Ladegerät, den Adapter und das Härtelicht separat in die Rücksendungsverpackung.
3) Senden Sie die Batterien gemäß den örtlichen Vorschriften.
Gewährleistung
Ultradent Products, Inc. ("Ultradent") garantiert, dass dieses Produkt für einen Zeitraum von 5 Jahren ab dem Kaufdatum, wenn es gemäß der dem Produkt beiliegenden
Bedienungsanleitung betrieben wird, (i) in allen wesentlichen Punkten den Spezifikationen entspricht, die in den dem Produkt beigefügten Unterlagen von Ultradent festgelegt sind;und (ii) frei von Material- und Verarbeitungsfehlern ist.
Diese beschränkte Garantie ist nicht übertragbar und gilt nur für den ursprünglichen Käufer und erstreckt sich nicht auf nachfolgende Eigentümer des Produkts. Diese beschränkte
Garantie gilt nicht für andere Zubehörkomponenten wie - aber nicht beschränkt auf - Batterien, Ladegeräte oder adaptive Linsen. Diese beschränkte Garantie erlischt, wenn das Produkt
aufgrund von Fahrlässigkeit, Missbrauch, Missbrauch, Unfall, Änderung, Manipulation, Änderung oder Nichtbeachtung der geltenden Gebrauchsanweisungen ausfällt oder beschädigt
wird. Nur zum Beispiel ist ein Produkt, das fallen gelassen und beschädigt wurde, nicht von dieser Garantie abgedeckt. Um unter diese beschränkte Garantie zu fallen, muss Ultradentzusammen mit dem defekten Produkt ein Kaufbeleg (z. B. Kaufbeleg oder ähnliche Unterlagen) vorgelegt werden.
Ein defektes Produkt, das die hierin festgelegten Garantiebedingungen erfüllt, wird nur nach dem Ermessen von Ultradent entweder repariert oder ersetzt. In keinem Fall übersteigt
die Haftung von Ultradent für das Produkt den vom Käufer gezahlten Kaufpreis. Unter keinen Umständen haftet Ultradent für indirekte, zufällige, vorhersehbare, unvorhergesehene,besondere oder Folgeschäden, die sich aus oder im Zusammenhang mit der Verwendung dieses Produkts ergeben.
*Mit Kaufbeleg mit Angabe des Verkaufsdatums an den Zahnarzt.

Instandhaltung

Allgemeine Reinigung des aushärtenden Lichts
Befeuchten Sie nach jedem Gebrauch eine Mull oder ein weiches Tuch mit einem zugelassenen Oberflächendesinfektionsmittel und wischen Sie die Oberfläche und die Linse ab. Nicht
zugelassene Reinigungsmittel können das Härtungslicht beschädigen.
ZULÄSSIGE REINIGUNGSMITTEL:• 70 % Isopropylalkohol• 70 % Ethanol

Allgemeine Reinigung des Lichtschutzes:
Reinigen Sie den VALO Kabellos-Lichtschutz mit einem beliebigen Oberflächendesinfektionsmittel. NICHT autoklavieren.
Reparatur
Vom Benutzer durchgeführte Wartung
1) Verwenden Sie eine Barrierehülse, um zu verhindern, dass Dentalkomposit an der Oberfläche der Linse haftet. Verwenden Sie bei Bedarf ein dental Instrument aus Kunststoff oder14

Bearbeitung

Dieses Produkt ist für die Verwendung mit einer Schutzhülle vorgesehen. Zur Wartung und allgemeine Reinigung siehe oben.

Lagerung und Entsorgung

Die Akkus immer entfernen, wenn die Polymerisationsleuchte über einen Zeitraum von mehr als 2 Wochen aufbewahrt oder für die Reise verpackt wird. Wenn Akkus für einen längerenZeitraum ohne Aufladen im Gerät verbleiben, können sie funktionsunfähig oder nicht aufladbar werden. Akkus nicht bei Temperaturen über 40 °C (104 °F) oder in direktem Sonnenlicht
aufbewahren.
Lagerung und Transport Polymerisationsleuchte:• Temperatur: +10 °C bis +40 °C (+50 °F bis +104 °F)• Relative Luftfeuchtigkeit: 10 % bis 95 %• Umgebungsdruck: 500 hPa bis 1060 hPa
15

Bei der Entsorgung von Elektronikschrott (z. B. Polymerisationsleuchten, Ladegeräte, Akkus und Netzteile) sind die örtlichen Abfall- und Recyclingrichtlinien zu befolgen.

Technische Gesichtspunkte
Zubehör

Effektive Wellenlängenbänder bei Composite-Aushärtung:

EFFEKTIVE VERBUNDHÄRTUNGSWELLENLÄNGENBÄNDER
(nutzbarer Wellenlängenbereich: 385-515 nm)

CE-Informationen
SPEKTRALE STRAHLUNGSLEISTUNG (mW/nm)

ArtikelTechnische Informationen / Daten

VALO Cordless Lichtschutzschirm

VALO Ladegerät

VALO Akkus

Effektiver niedrigerer
wellenlängenband
395-415 nm

Effektives oberes
Wellenlängenband
440-480 nm

Xtra
High-PowerStandard

WAVELENGTH (nm)MerkmalInformationen / Spezifikation

Linse

Durchmesser 11,7 mm

Wellenlängenbereich• Nutzbarer Wellenlängenbereich: 385 - 515 nm• Spitzenwert Wellenlängen: 395 - 415 nm und 440 - 480 nm
Vergleichstabelle Nennwert spezifische Ausstrahlung

Messinstrument

Tabelle Lichtstärke

16‡ Gigahertz-Spektrumsanalysator
Ausstrahlung

Gesamtleistung

Messgeräteöffnung

15 mm

15 mm

StandardLichtstärke(±10%)

900 mW/cm²

970 mW

Modus Hohe
Lichtstärke(±10%)

1500 mW/cm²

1615 mW

Maximal-Lichtstärke(±10%)

2100 mW/cm²

Die Strahlungsleistung variiert je nach
Leistungsfähigkeit des Geräts, Messmethodeund Lichtplatzierung.
‡ Radiant Exitance entspricht der ISO
10650, wenn es mit einem Gigahertz-Optik
Spektrumanalysator gemessen wird.

2260 mW

VALO Grand
Polymerisationsleuchte

Bewertungen: IEC 60601-1 (Sicherheit), IEC 606011-2 (EMV)

Gewicht:• Mit Akkus: 190 g (6 Unzen)• Ohne Akkus: 150 g (5 Unzen)
Abmessung: (203 × 32,5 × 27) mm (8 × 1,28 × 1,06) Zoll)

Netzteil Ladegerät

Ausgang - 12 VDC bei 500 mA
Eingang - 100 VAC bis 240 VAC
Ultradent Art.-Nr. 5930 VALO Netzteil Ladegerät mit
internationalen Steckereinsätzen

Bewertungen: IEC 60601-1 (Sicherheit)
Kabellänge - 1,8 Meter (6 Fuß)
Das Netzteil für das VALO-Ladegerät ist ein Netzteil der Klasse II für medizinische
Anwendungen und bietet eine Entkopplung von der STROMNETZVERSORGUNG

17

Sonstige Angaben

MerkmalInformationen / Spezifikation

VALO Ladegerät

VALO 3,6 VDC intelligentes Akkuladegerät für Lithium-Eisenphosphat-Akkus:• Automatische Abschaltung bei vollständiger Aufladung• Automatische Erkennung defekter Akkus• Schutzeinrichtungen: Thermisch, Überladung, Kurzschluss, Verpolungsschutz⸰ Rote LED - Aufladen⸰ Grüne LED - leer oder vollständig aufgeladen⸰ LED aus - Kurzschluss• Ladedauer: 1 - 3 StundenBewertung: CE, WEEE
Wiederaufladbar: Sichere Chemie Lithium-Eisenphosphat (LiFePO4) RCR123A• Betriebsspannung: 3,2 VDC
Bewertungen: UL, CE, RoHS, WEEEVALO Akkus

Temperatur: +10 °C bis +32 °C (+50 °F bis +90 °F)
Betriebs-bedingunRelative Luftfeuchtigkeit: 10 % bis 95 %gen
Umgebungsdruck: 700 hPa bis 1060 hPa
Arbeitszy-klus

Die Polymerisationsleuchte ist für den kurzzeitigen Betrieb ausgelegt. Bei maximaler Umgebungstemperatur (32 °C) fortlaufend 1 Minute
EIN und 30 Minuten AUS (Abkühlzeit).

Problembehandlung

Das Härtungslicht ist für den Einsatz in der unten angegebenen elektromagnetischen Umgebung bestimmt. Der Kunde oder Benutzer sollte sicherstellen, dass dieser
in einer solchen Umgebung verwendet wird.
Emissionsprüfung

Einhaltung

Elektromagnetische Umgebung - Orientierungshilfe

HF-Emissionen
CISPR 11

Gruppe 1

Das Härtungslicht verwendet Batterien und wird nicht durch EMI, RF oder Überspannungsschutz
beeinflusst.

HF-Emission
CISPR 11

Klasse B

Harmonische Emissionen
IEC 61000-3-2

Entfällt

Das Härtungslicht nutzt elektrische und elektromagnetische Energie nur für seine internen
Funktionen. Jedoch sind alle HF-Emissionen sehr gering und verursachen wahrscheinlich keine
Störungen bei elektronischen Geräten in der Nähe.

Emissionen durch
Spannungsschwankungen /Flimmern nach IEC 61000-3-3

Entfällt

Die Prüfung auf Oberwellenemission und Spannungsschwankungen ist für das Härtungslicht nicht
anwendbar, da es batteriebetrieben ist.
Das Härtungslicht ist für den Einsatz in allen Einrichtungen geeignet, auch in häuslichen
Einrichtungen und solchen, die direkt an das öffentliche Niederspannungsnetz angeschlossen
sind, das Gebäude für den häuslichen Gebrauch versorgt.

Leitfaden und Herstellungserklärung zur elektromagnetischen Störfestigkeit

Wenn mit den unten vorgeschlagenen Lösungen das Problem nicht behoben werden sollte, rufen Sie bitte Ultradent unter der Nr. 800.552.5512 an. Wenden Sie sichaußerhalb der USA an Ihren Ultradent-Händler oder Dentalhändler.

18

Leitfaden und Herstellungserklärung für elektromagnetische Emissionen

ProblemMögliche Lösungen

Die Lichtquelle wird nicht eingeschaltet

1) Drücken Sie auf die Taste Zeit/Modus oder Power, um aus dem Power-Sparmodus aufzuwachen.
2) Überprüfen Sie die rote Batterieladeanzeige auf den Ladezustand der Batterie.
3) Überprüfen Sie, ob die neuen Batterien richtig in das Gerät platziert sind.
4) Wenn die roten und gelben Warn-LEDs blinken, bedeutet dies, dass das Härtungslicht seinen internen
Temperatursicherheitsgrenzwert erreicht hat. Lassen Sie das Härtungslicht 10 Minuten lang abkühlen oder verwenden
Sie ein kühles, feuchtes Handtuch, um das Gerät schnell abzukühlen.
5) Wenn die rote Warn-LED blinkt und ständig piept, wenden Sie sich zur Reparatur an den Ultradent-Kundendienst.

Das Licht bleibt nicht für die
gewünschte Zeit an1) Überprüfen Sie Modus und Kontrollleuchten auf korrekte Zeiteingabe.
2) Überprüfen Sie die Batterieladeanzeige auf den Ladezustand der Batterie.
3) Überprüfen Sie, ob die neuen Batterien richtig in das Gerät platziert sind.

Keine ordnungsgemäße LichtAushärtung des Kunstharzes

1) Prüfen Sie die Linse auf Reste von ausgehärteten Harzen/Kompositen.
2) Vergewissern Sie sich mit dem richtigen gelben UV-Augenschutz, dass die LED-Leuchten funktionieren
3) Überprüfen Sie den Leistungspegel mit einem Lichtmesser. Bei Verwendung eines Lichtmessers empfiehlt Ultradent, das
Härtungslicht im Standard-Power-Modus zu überprüfen.
4) HINWEIS: Die tatsächliche numerische Leistung wird aufgrund der Ungenauigkeit üblichen Lichtmessgeräte und desspeziellen LED-Pakets im Lichthärtegerät verzerrt sein. Lichtmessgeräte unterscheiden sich stark und sind für bestimmte
Lichtleiterspitzen und -linsen konzipiert.
5) Verfallsdatum des aushärtenden Harzes überprüfen.
6) Stellen Sie sicher, dass die richtige Technik entsprechend den Empfehlungen des Herstellers befolgt wird.

Akkus werden nicht aufgeladen

1) Stellen Sie sicher, dass die Batterien in der richtigen Ausrichtung in das Ladegerät eingelegt sind und lassen Sie die
Batterien 1-3 Stunden lang aufladen.
2) Wenn die roten Lichter am Ladegerät nicht auf Grün wechseln, wenden Sie sich an den Ultradent-Kundendienst, um
Ersatzbatterien und/oder Ladegerät zu bestellen.
3) Wenn weder grüne noch rote Lichter am Ladegerät sichtbar sind, wenden Sie sich an den Ultradent-Kundendienst, um
das Ladegerät und/oder den Netzadapter zu bestellen oder zu ersetzen.

Akkus werden im Ladegerät nicht
aufgeladen

1) Stellen Sie sicher, dass das Ladegerät eingesteckt ist und das Netzteil an eine funktionierende Steckdose angeschlossen
ist.
2) Wenn grüne oder rote Lichter am Ladegerät nicht sichtbar sind, wenden Sie sich an den Ultradent-Kundendienst, um ein
neues Ladegerät und/oder ein neues Netzteil zu erhalten.

Modus oder Zeitintervalle können nicht
geändert werden

1) Halten Sie die Tasten Zeit/Modus und Ein/Aus gedrückt, bis eine Reihe von Pieptönen anzeigt, dass das Härtungslicht
entriegelt ist.

Das Härtungslicht ist für den Einsatz in der unten angegebenen elektromagnetischen Umgebung bestimmt. Der Kunde oder Benutzer sollte sicherstellen, dassdieser in einer solchen Umgebung verwendet wird.
Orientierungshilfe für die elektromagnetische
Umgebung

Störfestigkeits-prüfung

IEC 60601 Prüfpegel

Konformitätsstufe

Elektrostatische Entladung (ESD)±8 kV Kontakt±15 kV Luft

±8 kV Kontakt±15 kV Luft

Die physische Umgebung sollte auf Folgendes
beschränkt sein:
1) IP Code: IP20
2) Nicht in Flüssigkeit eintauchen.
3) Nicht in der Nähe von entflammbarem Gas
verwenden. Einheit ist nicht-APG und nicht-AP.
4) Luftfeuchtigkeitsbereich bei Lagerung: 10% - 95 %5) Temperaturbereich bei Lagerung: 10 °C - 40 °C

Schnelle vorübergehende /gehäufte Spannungsstöße
IEC 61000-4-4±2 kV fürStromversorgungsleitungen±1 kV für Eingangs-/
Ausgangsleitungen±2 kV fürStromversorgungsleitungen
Hinweis 1: Die
Polymerisationsleuchte hat keine
E/A-Ports

Die Polymerisationsleuchte wird mit Akkus
betrieben und kann nicht an das Wechselstromnetz
angeschlossen werden.

Überspannung
IEC 61000-4-5±1 kV Leitung gegen Leitung±2 kV Leitung gegen Erde

±1 kV Leitung gegen Leitung±2 kV Leitung gegen Erde

Voltage, dips, shorts, interruptions <5 % U
and variations on the power(>95 % Einbruch in U für 0,5
supply input lines
Zyklus)<5 % U(>95 % Einbruch in U für 0,5
Zyklus)IEC 61000-4-11

40 % U(60 % Einbruch in U für 5 Zyklen)

40 % U(60 % Einbruch in U für 5 Zyklen)

70 % U(30 % Einbruch in U für 25 Zyklen)

70 % U(30 % Einbruch in U für 25 Zyklen)<5 % U(>95 % Einbruch in U für 5 Sek.)<5 % U(>95 % Einbruch in U für 5 Sek.)
Hinweis 2: Eigenständige
Korrektur

IEC 61000-4-2

Da die Polymerisationsleuchte mit Akkus betrieben
wird, ist sie keinen elektrischen Transienten,Überspannungen, Spannungseinbrüchen,
Kurzschlüssen, Unterbrechungen oder Schwankungendes Wechselstromnetzes ausgesetzt.
Das Zubehör-Akkuladegerät kann den oben
genannten Bedingungen unterliegen, ist jedoch fürden Betrieb der VALO Cordless Polymerisationsleuchte
nicht unbedingt erforderlich.
Wenn die Batteriespannung der
Polymerisationsleuchte auf 4 VDC abfällt, ist der
Betrieb des Geräts nicht möglich. Die VALO CordlessPolymerisationsleuchte wird abgeschaltet. Wenn neue
Akkus eingesetzt wurden und die ordnungsgemäße
Lichtleistungspegel wiederhergestellt wurde, wird die
Polymerisationsleuchte neu gestartet und kehrt zurück
in den gleichen Zustand, der vor dem Stromausfall
bestand. Die Polymerisationsleuchte wird sich bei
Stromausfall wieder selbst einstellen.

19Leitfaden und Herstellungserklärung zur elektromagnetischen Störfestigkeit
Netzfrequenz(50/60 Hz) Magnetfeld

30 A/m30 A/mMagnetfelder mit Netzfrequenz sollten sich auf einem
Niveau befinden, das für einen typischen Standort für
das Gesundheitswesen in einer typischen häuslichen
Wohngegend, gewerblichen Gegend, Krankenhausoder Militärumfeld charakteristisch ist.

IEC 61000-4-8

Anleitung und Herstellungserklärung für empfohlene Trennungsabstände zwischen tragbaren und mobilen HF-Kommunikationsgeräten und der VALO CordlessPolymerisationsleuchte
Die Polymerisationsleuchte ist für den Einsatz in einer elektromagnetischen Umgebung vorgesehen, in der ausgestrahlte HF-Störungen kontrolliert werden. Der
Benutzer der Polymerisationsleuchte kann elektromagnetische Interferenzen verhindern, indem er wie unten empfohlen einen Mindestabstand zwischen tragbaren
und mobilen HF-Kommunikationsgeräten (Sendern) und der Polymerisationsleuchte gemäß der maximalen Ausgangsleistung des Kommunikationsgeräts einhält.
Trennungsabstand gemäß Frequenz des Senders(Meter)Hinweis 1: Die Polymerisationsleuchte ist nicht mit Ports oder zugänglichen E/A-Leitungen ausgestattet.

Anleitung und Herstellungserklärung zur elektromagnetischen Störfestigkeit für nicht-lebenserhaltende Systeme
Die Polymerisationsleuchte ist zur Verwendung in der unten angegebenen elektromagnetischen Umgebung vorgesehen. Der Kunde oder Benutzer sollte dafürsorgen, dass es in einer solchen Umgebung verwendet wird.
Störfestigkeitsprüfung

HF-Leitvermögen

IEC 61000-4-6

IEC 60601 Prüfpegel

3 Veff

150 kHz bis 80 MHz

Konformitätss-tufe

3 Veff

3 V/m

80 MHz – 800 MHz

800 MHz – 2,5 GHz

0,01

0,12 Meter

0,035 Meter

0,07 Meter0,10,37 Meter

0,11 Meter

0,22 Meter

Tragbare und mobile HF-Kommunikationsgeräte sollten nicht näher als der empfohlene
Trennungsabstand zu einem beliebigen Teil der Polymerisationsleuchte, einschließlich
der Kabel, verwendet werden, der sich aus der für die Frequenz des Senders geltenden
Gleichung berechnet.
Empfohlener Trennungsabstand:

1

1,7 Meter

0,35 Meter

0,7 Meter

10

3,7 Meter

1,11 Meter

2,22 Meter

100

11,7 Meter

3,5 Meter

7,0 Meter

150 kHz bis 80 MHz

3 V/m

150 kHz – 80 MHz

Orientierungshilfe für die elektromagnetische Umgebung

80 MHz bis 800 MHz

Ausgestrahlte HF

Maximale Ausgangsnennsleistungdes Senders(P in Watts)

Die Polymerisationsleuchte wurde gemäß IEC 60601-1-2: 2014 getestet und hat die Anforderungen mit einer ausgestrahlten Feldstärke von 10 V/m zwischen 80 MHz
und 2,5 GHz bestanden. In den obigen Formeln entspricht der Wert von 3 Vrms V1 und der Wert 10 V/m entspricht E1.
Bei Sendern, deren eingestufte maximale Ausgangsleistung nicht oben aufgeführt ist, kann der empfohlene Trennungsabstand d in Metern (m) mit der für die
Frequenz des Senders geltenden Gleichung geschätzt werden, wobei laut dem Senderhersteller P die maximale Ausgangsnennleistung des Senders in Watt (W) ist.
HINWEIS 1: Bei 80 MHz und 800 MHz gilt der Trennungsabstand für den höheren Frequenzbereich.
HINWEIS 2: Diese Richtlinien gelten möglicherweise nicht in allen Situationen. Die elektromagnetische Ausbreitung wird durch Absorption und Reflexion von
Aufbauten, Objekten und Personen beeinflusst.

800 MHz bis 2,5 GHz
P ist die maximale Ausgangsnennleistung des Senders in Watt (W) gemäß den Angaben
des Senderherstellers und d ist der empfohlene Trennungsabstand in Metern (m).
Die durch ein elektromagnetisches Standortgutachten ermittelte Feldstärken vonstationären HF-Sendern sollten in jedem Frequenzbereich unter der Konformitätsstufe
liegen.

IEC 61000-4-3

80 MHz bis 2,5 GHz

80 MHz bis 2,5 GHz

In der Nähe von Geräten, die mit dem folgenden Symbol gekennzeichnet sind, können
Interferenzen auftreten:

HINWEIS 1: Bei 80 MHz und 800 MHz gilt der höhere Frequenzbereich.
HINWEIS 2: Diese Richtlinien gelten möglicherweise nicht in allen Situationen. Die elektromagnetische Ausbreitung wird durch Absorption und Reflexion von
Aufbauten, Objekten und Personen beeinflusst.
aFeldstärken von fest installierten Sendern wie z. B. Basisstationen für Funktelefone (zellulare / cordlesse) und mobile Landfunkgeräte, Amateurfunk-, AM- und
FM-Radiosendungen und Fernsehsendungen können theoretisch nicht genau vorhergesagt werden. Zur Bewertung der elektromagnetischen Umgebung aufgrundvon stationären HF-Sendern sollte ein elektromagnetisches Standortgutachten in Betracht gezogen werden. Wenn die gemessene Feldstärke an dem Ort, an dem
die Polymerisationsleuchte verwendet wird, die oben angegebene zutreffende HF-Konformitätsstufe überschreitet, sollte die VALO Cordless Polymerisationsleuchte
beachtet werden, um den Normalbetrieb zu überprüfen. Wenn eine ungewöhnliche Leistung beobachtet wird, sind möglicherweise weitere Maßnahmen
erforderlich, wie z. B. eine Neuausrichtung oder ein Standortwechsel des Polymerisationsleuchte.
bÜber den Frequenzbereich von 150 kHz bis 80 MHz sollten die Feldstärken unter 3 V/m liegen.

20

21

VALO™ GRANDFRLAMPE A POLYMERISER
Description du produit

Avec son spectre à large bande, la lampe à polymériser VALO Grand est conçue pour polymériser tous les produits photopolymérisables dans la gamme de longueurs d'onde de385-515nm selon ISO 10650. La lampe à polymériser VALO Grand utilise les batteries rechargeables VALO et le chargeur de batterie d'Ultradent. La lampe à polymériser est conçue pours'insérer dans un support d'unit dentaire standard ou peut être montée sur mesure à l'aide du support fourni avec le kit.

Composants du produit VALO Grand :
•
•
•
•
•
•

Photo

Boutons et voyants

Ecran de protection sans fil VALO : Cordless
L'écran de protection VALO Cordless est de forme ovale, peut être tourné pour une utilisation maximale et peut être utilisé avec un manchon dr protection transparent.

LégendeBouton(s) d'allumage(haut ou bas)
Voyants de minuterie
Indicateur de pile faible
Voyant de mode/statut
Bouton de changement detemps/mode(bas)(haut)For all products described, carefully read and understand all instructions and SDS information prior to use.

Indications d'utilisation/Utilisation prévue

La source de lumière pour le traitement des matériaux de restauration dentaire et des adhésifs photo-polymérisés.

Avertissements et précautions
Groupe de risque 2
ATTENTION UV émis par ce produit. Une irritation des yeux ou de la peau peut résulter de l'exposition. Utilisez un écran de protection contre la lumière approprié.
ATTENTION Risque possible de radiation optique émise par ce produit. Ne regardez pas la lampe lors de son fonctionnement. Peut être nocif pour les yeux..
• Ne regardez pas directement la lumière émise par la lampe. Le patient, le clinicien et les assistants doivent toujours porter une protection oculaire UV de couleur ambre lorsque lalampe à polymériser est utilisée.
• Pour éviter tout risque d'électrocution, il est interdit de modifier cet appareil. Utilisez uniquement le bloc d'alimentation et les adaptateurs de prise Ultradent VALO fournis. Si ces
composants sont endommagés, ne les utilise pas et appelle le service clientèle d'Ultradent pour commander un remplacement.
• Les équipements de communication RF portables peuvent dégrader les performances s'ils sont utilisés à moins de 30 cm.
• N'utilisez que des accessoires, des câbles et des blocs d'alimentation autorisés afin d'éviter un mauvais fonctionnement, une augmentation des émissions électromagnétiques ou une
diminution de l'immunité électromagnétique (voir la section Émissions électromagnétiques).
• Pour éviter les risques d'incendie électrique liés à la manipulation des piles :⸰ Ne mélangez pas des piles rechargeables avec des piles non rechargeables ou d'autres types de piles.
⸰ N'essayez PAS de charger des piles non rechargeables. Charger des piles non rechargeables peut entraîner des blessures ou des dommages matériels.
⸰ NE PAS autoclaver ou pulvériser les piles, les contacts des piles, le chargeur ou l'adaptateur d'alimentation CA avec un liquide de quelque nature que ce soit. Si de la corrosion
apparaît sur les contacts du chargeur de batterie, appelez le service clientèle d'Ultradent pour commander un remplacement.
⸰ Ne chargez pas les batteries à proximité de matériaux inflammables.
⸰ Ne gardez pas le chargeur dans la salle d'opération clinique.
• Pour éviter tout risque de blessure, NE PAS utiliser de piles corrodées (rouille), bosselées, dégageant une odeur ou des fluides, dont l'emballage est déchiré ou manquant, ou qui
sont endommagées de quelque manière que ce soit. Appelez le service clientèle d'Ultradent pour commander des piles de remplacement.
• N'utilisez que des piles recommandées. Les piles alternatives peuvent entraîner des dysfonctionnements.
• Pour éviter tout risque d'irritation ou de blessure thermique, évitez les cycles de polymérisation consécutifs et n'exposez pas les tissus mous buccaux à proximité pendant plus de 10
secondes, quel que soit le mode. Si des temps de polymérisation plus longs sont nécessaires, utilisez plusieurs cycles de polymérisation avec des périodes de repos entre les cycles,
ou utilisez un produit à double polymérisation pour éviter de surchauffer les tissus mous.
• Soyez prudent lorsque vous traitez des patients qui souffrent de réactions photobiologiques indésirables ou de sensibilités, des patients qui suivent un traitement de chimiothérapie
ou des patients traités avec des médicaments photosensibilisants.
• Cet appareil peut être sensible à de forts champs magnétiques ou électriques statiques, qui pourraient perturber la programmation. Si vous pensez que cela s'est produit,
débranchez l'appareil momentanément, puis rebranchez-le dans la prise de courant.
• NE PAS essuyer la lampe à polymériser avec des nettoyants caustiques ou abrasifs, ne pas passer à l'autoclave ou l'immerger dans un bain ultrasonique, un désinfectant, une solution
22

Instructions par étapes
Préparation

1) Chargez les piles avant d'utiliser la lampe à polymériser (voir la section Entretien des piles).
2) Avant chaque utilisation, placez un nouveau manchon de protection sur la lampe à polymériser, en minimisant les plis sur la lentille pour obtenir les meilleurs résultats.
Note :• Pour éviter la contamination croisée et empêcher le matériau composite dentaire d'adhérer à la surface de la lentille et du corps de la baguette, un manchon de protection approuvépar Ultradent doit être utilisé sur la lampe à polymériser VALO à chaque utilisation. Les manchons de protection sont destinées à être utilisées pour un seul patient.

1 - Lampe à polymériser VALO Grand4 - Piles rechargeables Ultradent VALO
1 - Chargeur de batterie Ultradent VALO avec adaptateur de courant alternatif 12VDC de qualité médicale
1 - Échantillon de manchons de protection VALO
1 - Ecran de protection VALO Grand
1 - Support de montage de la lampe de polymérisation avec ruban adhésif double face

Vue générale des contrôles :de nettoyage ou un liquide quelconque. Le non-respect des instructions de traitement incluses peut rendre la lampe à polymériser inopérante.
• Pour éviter d'endommager l'équipement, N'INSÉREZ PAS vos doigts, des instruments ou d'autres objets dans le compartiment à piles de la lampe à polymériser.
• Pour éviter d'endommager l'équipement, n'essayez PAS de nettoyer les contacts dorés ou toute autre partie du compartiment de la batterie. Appelez le service clientèle d'Ultradenten cas de problème.
• Pour éviter tout risque de contamination croisée, les manchons de protection sont à usage unique.
• Pour réduire le risque de corrosion, retirez le manchon de protection après utilisation.
• Pour réduire le risque de sous-polymérisation des résines, n'utilisez pas la lampe à polymériser si la lentille est endommagée.

Utilisation

1) Chaque mode d'alimentation est utilisé pour la polymérisation de matériaux dentaires avec des photo-initiateurs. Consultez le Guide de mode rapide pour connaître les temps depolymérisation recommandés.
REMARQUE : La lampe à polymériser est programmée pour passer successivement du mode Standard Power au mode High-Power Plus et au mode Xtra Power. Par exemple, pourpasser du mode Standard Power au mode Xtra Power, il est nécessaire de passer en mode High-Power Plus, puis en mode Xtra Power.
2) La lampe à polymériser mémorise l'intervalle de temps et le mode les plus récemment utilisés, et revient par défaut à ces derniers chaque fois que les modes sont modifiés ou queles piles sont retirées.

Fonctionnement
MODE DE POLYMÉRISATION : Mode de puissance standard

INTERVALLES DE PROGRAMMATION : 5, 10, 15 et 20 secondes.
• La lampe à polymériser passe par défaut à ce mode lorsqu’elle est initialement allumée. Le voyant de Mode/Statut sera vert et les quatre voyants de minuterie verts sont allumés,
indiquant le mode d'alimentation standard.
• Pour modifier les intervalles de programmation de minuterie, appuyez sur le bouton Temps/Mode.
• Appuyez sur le bouton d’allumage pour polymériser. Pour arrêter la polymérisation avant la fin d’un intervalle de minuterie, appuyez à nouveau sur le bouton d’allumage.

MODE DE POLYMÉRISATION : Mode haute puissance plus

INTERVALLES DE PROGRAMMATION : 1, 2, 3 et 4 secondes.
• En mode de puissance standard, maintenez enfoncé le bouton de changement de Temps/Mode pendant 2 secondes, puis relâchez-le. Le voyant de Mode/Statut sera orange et lesquatre voyants de minuterie verts s'allumeront et clignoteront, indiquant le mode Haute puissance.
• Pour modifier les intervalles de programmation de minuterie, appuyez sur le bouton Temps/Mode.
• Appuyez sur l'un des boutons d'allumage pour polymériser. Pour arrêter la polymérisation avant la fin d'un intervalle de minuterie, appuyez à nouveau sur l'un des boutons
d'allumage.
• Pour revenir au mode de puissance standard, maintenez enfoncé le bouton de changement de Temps/Mode pendant 2 secondes, puis relâchez-le pour passer en mode Puissanceextra. Appuyez et maintenez à nouveau pendant 2 secondes, puis relâchez. Le voyant de Mode/Statut sera vert et les quatre voyants de minuterie verts sont allumés, indiquant lemode Standard.

MODE DE POLYMÉRISATION : Mode de puissance extra

INTERVALLE DE PROGRAMMATION : 3 secondes seulement (Remarque : Le mode Puissance extra comporte un délai de sécurité de 2 secondes à la fin de chaque cycle depolymérisation afin de limiter le chauffage pendant le polymérisation consécutif. A la fin du délai, un bip sonore indique que l'unité est prête à l'utilisation).
• En mode Puissance standard, appuyez sur le bouton de changement de Temps/Mode pendant 2 secondes, relâchez, maintenez enfoncé pendant 2 secondes, puis relâchez. Le
voyant de Mode/Statut sera orange et clignotera, et trois des voyants de minuterie verts s'allument et clignotent, indiquant le mode Puissance extra.
• Appuyez sur le bouton d’allumage pour polymériser. Pour arrêter la polymérisation avant la fin d’un intervalle de programmation de minuterie, appuyez à nouveau sur le boutond’allumage.
• Pour revenir au mode Puissance standard, maintenez le bouton Temps/Mode enfoncé pendant 2 secondes, puis relâchez-le. Le voyant de Mode/statut sera vert et les voyants deminuterie verts sont allumés, indiquant le mode Puissance standard.

Mode veille : La lampe à polymériser passe en mode veille après 60 secondes d'inactivité, comme l'indique le clignotement lent du voyant de mode/statut. Le fait de toucher ou de saisir
l'unité activera la lampe à polymériser et la ramènera automatiquement au dernier réglage utilisé. Pour maximiser la durée de vie des piles, laissez la lampe de polymérisation immobile
lorsqu'elle n'est pas utilisée.

Nettoyer1) Jeter les manchons de protection usagés avec les déchets standards après chaque patient.
2) Voir la section Traitement.

Instructions du support de montage

1) Le support doit être monté sur une surface plane et non grasse.
2) Nettoyez la surface avec de l'alcool.
3) Décollez le ruban adhésif du support.
4) Positionnez le support de sorte que la lampe à polymériser puisse être soulèvée lorsqu'elle est retirée. Appuyez fermement pour la faire tenir en place.

Guide rapide des modes :23

Puissance standard
1000 mW/cm2

Mode

Haute puissance plus
1600 mW/cm2

Puissance extra
3200 mW/cm2

Bouton d’allumage
LED de mode/de temps

Boutons de temps

Options de temps

5s

10s

15s

20s

1s

2s

3s

4s

3s seulement

Pour changer le temps

Appuyez et relâchez rapidement le bouton Temps pour passer d'une durée à l'autre.

Pour changer de mode

Appuyez sur le bouton Temps et maintenez-le enfoncé pendant 2 secondes, puis relâchez-le. La lampe à polymériser passera
au mode suivant.

LégendeLED fixes

LED clignotantes

Guide rapide de polymérisation :Mode de polymérisation

Puissance* (mW)

Irradiance* (mW/cm2)

Temps d'exposition total (secondes)

Énergie** (Joules)

Standard

970

900

5/10/15/20

4.8J/9.7J/14.5J/19.4J

High Power Plus

1,620

1,500

1/2/3/4

1.6J/3.2J/4.8J/6.5J

Xtra

2,260

2,100

3

6.8J

*Sortie nominale à une distance de 2 mm entre l'extrémité de la lentille et la surface supérieure du composite.
**Les valeurs de la section énergie totale par cycle (joules) sont arrondies au dixième le plus proche.

Guide rapide des avertissement :Avertissement relatif au niveau depuissance

Avertissement relatif à la température

Avertissement relatif à l'entretien

Avertissement relatif aux LED

Remplacez les piles

Permettre la récupération

Appeler le service clientèle pourréparation

Appeler le service clientèle pourréparation• Piles déchargées : clignotement
lent• Arrêt : 3 bips, clignotement• Utilisation interdite• 3 bips• Clignotement lent
• Utilisation interdite• Pas de son• Clignotement, 2 secondes
d’intervalle• Utilisation autorisée• 3 bips continus• Un clignotement rapide• Utilisation interdite

Réparation
Entretien effectué par l'utilisateur
1) Utilisez un manchon de protection pour empêcher le composite dentaire d'adhérer à la surface de la lentille. Si nécessaire, utilisez un instrument dentaire en plastique ou en acierinoxydable pour retirer avec précaution tout composite collé. N'utilisez pas d'outils susceptibles d'endommager la lentille.
2) Les radiomètres sont très différents les uns des autres et sont conçus pour des embouts et des lentilles spécifiques. Ultradent recommande de vérifier régulièrement la puissancesortie en mode Standard Power. REMARQUE : le résultat numérique réel sera faussé en raison de l'imprécision des radiomètres courants et du pack de LED personnalisé dans la lampe à
polymériser
Entretien des piles
Chargement et remplacement des piles
La lampe à polymériser est livrée avec 4 piles rechargeables au lithium-phosphate de fer. Ultradent recommande d'utiliser uniquement les piles rechargeables VALO.
Comment charger les piles :1) Placez le chargeur et les piles dans un endroit où ils n'entreront pas en contact avec des liquides ou des sprays désinfectants.
2) Insérez les piles dans le chargeur en orientant l'extrémité positive (+) vers les voyants lumineux du chargeur.
3) Les lumières vertes indiquent que les piles sont prêtes à l'emploi.
4) Le chargement des piles prend de 1 à 3 heures. Laissez les piles dans le chargeur jusqu'à ce qu'elles soient prêtes à l'emploi.
REMARQUE : Si le voyant rouge du chargeur ne passe pas au vert lorsque les piles ont été chargées pendant plus de trois heures, il se peut que la pile soit dégradée (corrompue) et qu'elle
ne puisse pas être chargée. Essayez une nouvelle pile ou appelez le service clientèle d'Ultradent pour commander un nouveau jeu de piles rechargeables.
Comment changer/insérer les piles :1) Retirez le capuchon arrière en le tournant d'un quart de tour dans le sens inverse des aiguilles d'une montre.
2) Retirez les piles.
3) Insérez les piles neuves en commençant par le côté positif (+).
4) Remettez le capuchon arrière en place en l'alignant et en le poussant doucement tout en le tournant dans le sens des aiguilles d'une montre. Le capuchon s'enclenche lorsqu'il est
complètement fixé.
5) L'appareil est prêt à l'emploi.
Piles faibles :La lampe à polymériser signale à l'utilisateur qu'il est temps de changer les piles lorsque le voyant de piles faibles clignote en rouge. Si la charge des piles devient trop faible,
un avertissement sonore de 3 bips se produit et la lampe à polymériser ne peut plus fonctionner jusqu'à ce que les piles soient rechargées ou que des piles neuves soient insérées. (Voirle guide rapide d'avertissement)
Réparation par le fabricant
1) Les réparations ne doivent être effectuées que par du personnel de service autorisé. Ultradent doit fournir au personnel d'entretien la documentation nécessaire pour effectuer lesréparations.
2) Lorsque vous envoyez des appareils pour réparation, entretien ou étalonnage, retirez toujours les piles de la lampe à polymériser et du chargeur. Emballez les piles, le chargeur,l'adaptateur et la lampe de polymérisation séparément dans la boîte de retour.
3) Expédiez les batteries conformément aux réglementations locales.

Garantie

Ultradent Products, Inc. ("Ultradent") garantit que ce produit, pendant une période de 5 ans à compter de la date d'achat, lorsqu'il est utilisé conformément au mode d'emploi fourni avec
le produit, (i) est conforme à tous égards matériels aux spécifications énoncées dans la documentation d'Ultradent accompagnant le produit ; et (ii) est exempt de défauts matériels et defabrication.
Cette garantie limitée n'est pas transférable et s'applique uniquement à l'acheteur d'origine et ne s'étend pas aux propriétaires ultérieurs du produit. Cette garantie limitée ne couvre pas
les autres composants accessoires tels que, mais sans s'y limiter, les piles, les chargeurs, les adaptateurs ou les lentilles adaptatives. Cette garantie limitée est annulée si le produit tombeen panne ou est endommagé en raison d'une négligence, d'un abus, d'une mauvaise utilisation, d'un accident, d'une modification, d'une altération ou d'un non-respect du mode d'emploi
applicable. À titre d'exemple uniquement, un produit qui tombe et qui est endommagé n'est pas couvert par cette garantie. Pour bénéficier de cette garantie limitée, la preuve d'achat (parexemple, le ticket de caisse ou un document similaire) doit être soumise à Ultradent en même temps que le produit défectueux.
Un produit défectueux répondant aux conditions de garantie énoncées dans le présent document sera, à la seule discrétion d'Ultradent, réparé ou remplacé. En aucun cas, la
responsabilité d'Ultradent pour le produit ne dépassera le prix d'achat payé par l'acheteur. En aucun cas, Ultradent ne sera responsable des dommages indirects, accessoires, prévus,imprévus, spéciaux ou consécutifs découlant de l'utilisation de ce produit ou en rapport avec celle-ci.

Traitement

Ce produit est destiné à être utilisé avec un manchon de protection. Voir ci-dessus pour l'entretien et le nettoyage général.
**Son utilisation peut entraîner une décoloration

Stockage et élimination

Si vous stockez la lampe à polymériser pendant plus de deux semaines ou si vous l’emportez en déplacement, enlevez toujours les piles. Si les piles demeurent dans l'appareil pendantde longues périodes sans être rechargées, elles peuvent devenir inutilisables ou impossibles à charger. N'entreposez pas les piles à des températures supérieures à 40 °C (104 °F) ou àla lumière directe du soleil.

Entretien
Réparation

Nettoyage général de la lampe à polymériser
Après chaque utilisation, humidifiez une gaze ou un chiffon doux avec un désinfectant de surface approuvé et essuyez la surface et la lentille. Les nettoyants non autorisés peuvent
endommager la lampe à polymériser.
LES NETTOYANTS ACCEPTES :• Alcool isopropylique à 70%• 70% d'éthanol
Nettoyage général de l'écran de protection :Nettoyez l'écran de protection VALO Cordless à l'aide d'un désinfectant de surface. Ne pas stériliser à l'autoclave.
24

Curing light Storage and Transport:• Température : +10 °C à +40 °C (+50 °F à +104 °F)• Humidité relative : 10 % à 95 %• Pression ambiante : 500 hPa à 1060 hPa
Lors de la mise au rebut des déchets électroniques (c'est-à-dire des lampes à polymériser, des chargeurs, des piles et des blocs d'alimentation), suivez les directives locales en matière dedéchets et de recyclage.

Considérations techniques
Accessoires

25Article

Informations CE

Informations/données techniques

Plages de longueur d'onde de polymérisation efficace des composites :Écran de protection contre la lumière
VALO sans fil

BANDES DE LONGUEURS D'ONDE EFFICACES POUR LE DURCISSEMENT DESCOMPOSITES(Gamme de longueurs d'onde utilisables : 385-515 nm)
PUISSANCE RADIANTE SPÉCIALE (mW/nm)

Chargeur VALO

Piles VALO

Efficace plus bas
bande de longueur
d'onde395-415 nm

Bande de longueur
d'onde supérieure
effective
440-480 nm

XtraHigh PowerStandard

WAVELENGTH (nm)Attribut

Informations/spécifications

LentilleDiamètre 11,7 mm

Plage de longueur
d'onde• Plage de longueur d'onde utilisables : 385 - 515nm• Longueurs d'onde maximale : 395 - 415nm et 440 - 480nm
Tableau comparatif d'exitance énergétique nominale
Instrument demesureTableau d'intensité
lumineuse

26‡ Analyseur de spectre Gigahertz
Exitance

Puissance totale

Ouverture del'appareil demesure

15 mm

15 mm

Puissance standard(±10%)

900 mW/cm²

970 mW

Haute puissanceplus (±10%)1500 mW/cm²

1615 mW

Puissance extra(±10%)

2100 mW/cm²

2260 mW

L'exitance radiante varie en fonction de lacapacité de l'instrument, de la méthode demesure et de l'emplacement de la lumière.
‡ L'exitance radiante est conforme à la norme
ISO 10650 lorsqu'elle est mesurée à l'aide d'unanalyseur de spectre Gigahertz-Optik.

Lampe àpolymériser VALO
Grand

Certifications : CEI 60601-1 (Sécurité), CEI
60601-1-2 (CEM)

Poids :• Piles comprises : 6 oz (190 grammes)• Sans piles : 5 oz (150 grammes)
Dimensions : (8 x 1,28 x 1,06) pouces, (203 x 32,5 x 27) mm

Chargeur d'alimentation

Puissance fournie - 12V CC à 500mA
Puissance absorbée - 100V CA à 240VCA
Alimentation pour chargeur Ultradent P/N 5930
VALO avec prises internationales

Certifications : IEC 60601-1 (Sécurité)Longueur du cordon - 1,8 m (6 ft)
L'alimentation du chargeur VALO est une alimentation de classe II dequalité médicale et fournit une isolation de l'alimentation secteur.

27

Attribut

Informations/spécifications

Directives et déclaration du fabricant en matière d'émissions électromagnétiques
La lampe à polymériser est destinée à être utilisée dans l'environnement électromagnétique spécifié ci-dessous. Le client ou l'utilisateur de l'appareil doit veiller à cequ'il soit utilisé dans ce type d'environnement.

Chargeur VALO

Chargeur intelligent de piles lithium au phosphate de fer 3,6 V CC VALO :• Arrêt automatique une fois les piles chargées• Détection automatique des piles défectueuses• Protections : Thermique, surcharge, court-circuit, inversion de polarité
⸰ LED rouge – charge
⸰ LED verte – vide ou entièrement chargé⸰ LED éteinte – court-circuit• Temps de charge : 1 à 3 heures
Certifications : CE, WEEE

Piles VALO

Rechargeables : Safe chemistry lithium au phosphate de fer (LiFePO4) RCR123A• Tension de travail : 3,2 V CC
Certifications : UL, CE, RoHS, DEEE

Conditions defonction-nement

Température : +10 °C à +32 °C (+50 °F à +90 °F)
Humidité relative : 10 % à 95 %Pression ambiante : 700 hPa à 1060 hPa

Cycle de service

La lampe à polymériser est conçue pour une utilisation à court terme. À la température ambiante maximale (32 °C), 1 minuteconsécutive d'utilisation, 30 minutes d'arrêt (période de refroidissement).

Dépannage
Si les solutions proposées ci-dessous ne rectifient pas le problème, appeller Ultradent au 800.552.5512. En dehors des États-Unis, appeller votre distributeur Ultradentou votre revendeur de produits dentaires.
Problème

La lumière ne s'allume pas1) Appuyez sur le bouton de changement de mode et de durée d'illumination ou le bouton Alimentation pour sortir dumode économie d'énergie.
2) Vérifiez l'état de charge de la pile à l'aide de l'indicateur rouge de pile faible.
3) Vérifiez que des piles neuves sont correctement insérées dans l'appareil.
4) Si les LED d'avertissement rouge et jaune clignotent, cela signifie que la lampe à polymériser a atteint la limite desécurité de sa température interne. Laissez la lampe à polymériser refroidir pendant 10 minutes ou utilisez une serviettehumide et froide pour refroidir l'appareil rapidement.
5) Si le voyant d'avertissement rouge clignote et émet des bips continus, appelez le service clientèle d'Ultradent pourréparation.

La lumière ne reste pas allumée
pendant la durée souhaitée

1) Vérifiez les voyants Mode et Durée pour une saisie correcte de la durée d'illumination.
2) Vérifiez l'état de charge de la pile à l'aide de l'indicateur de pile faible.
3) Vérifiez que des piles neuves sont correctement insérées dans l'appareil..

La lumière ne polymérise pas
correctement les résines

1) Vérifiez que la lentille ne présente pas de résidus de résines/composites polymérisés.
2) En utilisant une protection oculaire ambre UV appropriée, vérifiez que les lumières LED fonctionnent.
3) Vérifiez le niveau de puissance à l'aide d'un posemètre. Si vous utilisez un radiomètre, Ultradent recommande de vérifier
la lampe à polymériser en mode Standard Power.
REMARQUE : le résultat numérique réel sera faussé en raison de l'imprécision des radiomètres courants et du pack de LED
personnalisé utilisé par la lampe à polymériser. Les radiomètres sont très différents les uns des autres et sont conçus pourdes embouts et des lentilles spécifiques.
4) Vérifiez la date de péremption de la résine de polymérisation.
5) Assurez-vous que la technique utilisée est conforme aux recommandations du fabricant.

Les piles ne se rechargent pas

1) Assurez-vous que les piles sont insérées dans le chargeur dans le bon sens et laissez les piles se charger pendant 1 à
3 heures.
2) Si les voyants rouges du chargeur ne passent pas au vert, appelez le service clientèle d'Ultradent pour commander despiles et/ou un chargeur de remplacement.
3) Si aucun voyant vert ou rouge n'est visible sur le chargeur, appelez le service clientèle d'Ultradent pour commander ou
remplacer le chargeur et/ou l'adaptateur secteur.

Le chargeur ne charge pas les piles

1) Assurez-vous que le chargeur est branché et que l'adaptateur CA est branché sur une prise de courant qui fonctionne.
2) Si les lumières vertes ou rouges du chargeur ne sont pas visibles, appelez le service clientèle d'Ultradent pour obtenir unnouveau chargeur et/ou un nouvel adaptateur CA.

Impossible de changer de mode ou dedurée d'illumination

1) Maintenez les deux boutons Time/Mode et Power enfoncés jusqu'à ce qu'une série de bips indique que la lampe à
polymériser est déverrouillée.

Informations diverses
28

Solutions possibles

Test d'émission

Conformité

Directive sur l'environnement électromagnétique

Émissions RF
CISPR 11

Groupe 1

La lampe à polymériser utilise des piles et n'est pas affectée par les interférencesélectromagnétiques, les radiofréquences ou la suppression des surtensions.

Émissions RF
CISPR 11Classe B

Émissions harmoniques
CEI 61000-3-2

N/A

La lampe à polymériser utilise l'énergie électrique et électromagnétique uniquement pour ses
fonctions internes. Cependant, les émissions de radiofréquences sont très faibles et ne risquentpas de provoquer des interférences avec les équipements électroniques situés à proximité.

Fluctuations de tension/émissions de scintillement CEI
61000-3-3

N/A

Les tests d'émission d'harmoniques et de fluctuation de tension ne s'appliquent pas à la lampe à
polymériser car elle est alimentée par une pile.
La lampe à polymériser peut être utilisée dans tous les établissements, y compris lesétablissements domestiques et ceux qui sont directement raccordés au réseau publicd'alimentation électrique à basse tension qui alimente les bâtiments à usage domestique.

Directives et déclaration du fabricant en matière d'immunité électromagnétique
La lampe à polymériser est destinée à être utilisée dans l'environnement électromagnétique spécifié ci-dessous. Le client ou l'utilisateur de l'appareil doit veiller à cequ'il soit utilisé dans ce type d'environnement.
Test d'IMMUNITÉ

Niveau de test CEI 60601

Niveau de conformité

Directive sur l'environnement électromagnétique

Décharges électrostatiques (ESD)

Contact ± 8 kV
Air ± 15 kV

Contact ± 8 kV
Air ± 15 kV

L'environnement physique doit être restreint
comme suit ::1) Code IP : IP20
2) Ne pas immerger le produit dans un liquide.
3) Ne pas utiliser à proximité de gaz inflammable.
L'appareil n'appartient pas à la catégorie AP
ou APG.
4) Plage d'humidité de stockage : 10 % à 95 %5) Plage de température de stockage : 10 °C à 40 °C

Transitoires électriques rapides/
en salves
CEI 61000-4-4± 2 kV pour les lignes
d'alimentation± 1 kV pour les lignes d'entrée/sortie± 2 kV pour les lignes
d'alimentation
Remarque 1 : la lampe à
polymériser n'a pas de ports
d'entrée/sortie

La lampe à polymériser est alimentée par piles et nepeut pas être connectée à une alimentation secteur.

Pic
CEI 61000-4-5± 1 kV de ligne à ligne± 2 kV d'une ligne à la terre± 1 kV de ligne à ligne± 2 kV d'une ligne à la terre

Creux de tension, coupure
de courant et fluctuations dela tension de l'alimentationélectrique<5 % U(>95 % de baisse de tension en U
pendant 0,5 cycle)<5 % U(>95 % de baisse de tension en U
pendant 0,5 cycle)

40 % U(60 % de baisse de tension en U
pendant 5 cycles)

40 % U(60 % de baisse de tension en U
pendant 5 cycles)

70 % U(30 % de baisse de tension en U
pendant 25 cycles)

70 % U(30 % de baisse de tension en U
pendant 25 cycles)<5 % U(>95 % de baisse de tension
pendant 5 s)

<5 % U(>95 % de baisse de tension
pendant 5 s)
Remarque 2 :Récupérations automatiques

30 A/m30 A/m

CEI 61000-4-2

CEI 61000-4-11

Fréquence d'alimentation
Champ magnétique (50/60 Hz)
CEI 61000-4-8

Étant donné que la lampe à polymériser est alimentéepar piles, elle n’est pas soumise aux variationsélectriques, aux surtensions, aux creux de tension, auxcourts-circuits, aux interruptions ou aux variations del’alimentation secteur.
Le chargeur de piles auxiliaire peut être sujet aux
facteurs précédents, mais il est séparé et non essentiel
pour le fonctionnement de la lampe à polymériser
VALO sans fil.
Si la tension de la lampe à photopolymériser est
inférieure à 4 V CC, l'appareil ne fonctionnera plus. La
lampe à polymériser VALO sans fil s'éteindra. Lorsquede nouvelles piles sont installées et que les niveauxde puissance appropriés sont restauré, la lampe à
polymériser redémarre et revient au même état utilisé
avant la coupure de courant. La lampe à polymériser
se régénère automatiquement en cas de coupure
de courant.

Les champs magnétiques de fréquence d'alimentation
doivent être à des niveaux caractéristiques d'unemplacement typique dans un environnement
typique, résidentiel, de soins de santé à domicile,commercial, hospitalier ou militaire.

29

Directives et déclaration du fabricant en matière d'immunité électromagnétique
Remarque 1 : La lampe à polymériser n'est équipée d'aucun port ni d'aucune ligne d'E/S accessible.

Directives et déclaration du fabricant en matière d'immunité électromagnétique des systèmes d'assistance non vitaux
La lampe à polymériser est destinée à être utilisée dans l'environnement électromagnétique spécifié ci-dessous. Le client ou l'utilisateur de l'appareil doit veiller à cequ'il soit utilisé dans ce type d'environnement.
Test d'IMMUNITÉ

Émissions RF parconduction

CEI 61000-4-6

Niveau de test CEI
60601

3 Vrms150 kHz à 80 MHz

Niveau de conformité

3 Vrms150 kHz à 80 MHz

Directive sur l'environnement électromagnétiqueLes équipements de communication RF portables et mobiles ne doivent pas être utilisés
plus près d'une partie de la lampe à polymériser, y compris les câbles, que la distancede séparation recommandée calculée à partir de l'équation applicable à la fréquence del'émetteur.
Distance de séparation recommandée :80 MHz à 800 MHz

800 MHz à 2,5 GHz
Émissions RF parrayonnement

3 V/m

3 V/m

P est la puissance nominale maximale de sortie du transmetteur exprimée en watts (W)indiquée par le fabricant de l'émetteur et d est la distance de séparation recommandée
en mètres (m).
Les forces de champ des émetteurs RF fixes, telles que déterminées par un siteélectromagnétiquea, doivent être inférieures au niveau de conformité de chaque gammede fréquencesb.

CEI 61000-4-3

80 MHz à 2,5 GHz

80 MHz à 2,5 GHz

Directives et déclaration du fabricant concernant les distances de séparation recommandées entre les équipements de communication RF portables et mobiles et lalampe à polymériser sans VALO sans fil
La lampe à polymériser est destinée à être utilisée dans un environnement électromagnétique dans lequel les perturbations RF rayonnées sont contrôlées.
L'utilisateur de la lampe à polymériser peut aider à prévenir les interférences électromagnétiques en maintenant une distance minimale entre les équipements
de communication RF portables et mobiles (émetteurs) et la lampe à polymériser recommandée ci-dessous, en fonction de la puissance de sortie maximale deséquipements de communication.
Distance de séparation en fonction de la fréquence de l'émetteur(en mètres)
Puissance maximale de sortie nominale
de l'émetteur(P en Watts)

150 kHz à 80 MHz

80 MHz à 800 MHz

800 MHz à 2,5 GHz

0,01

0,12 mètre

0,035 mètres

0,07 mètres

0,10,37 mètres

0,11 mètres

0,22 mètres

1

1,7 mètres

0,35 mètres

0,7 mètres

10

3,7 mètres

1,11 mètres

2,22 mètres

100

11,7 mètres

3,5 mètres

7,0 mètres

La lampe à polymériser a été testée conformément à la norme CEI 60601-1-2:2014 et soumise à des champs de rayonnement de 10 V/m compris entre 80 MHz et 2,5GHz. La valeur de 3Vrms correspond à V1 et la valeur 10 V/m correspond à E1 dans les formules ci-dessus.
Pour les émetteurs dont la puissance maximale nominale ne figure pas ci-dessus, la distance de séparation recommandée d en mètres (m) peut être estimée àl'aide de l'équation applicable à la fréquence de l'émetteur, où P est la puissance nominale maximale de sortie de l'émetteur exprimée en watts (W) indiquée par lefabricant de l'émetteur.
REMARQUE 1 À 80 MHz et 800 MHz, la distance de séparation pour la plage de fréquences la plus haute s'applique.
REMARQUE 2 : ces directives ne s'appliquent pas nécessairement dans tous les cas. La propagation électromagnétique est influencée par l'absorption et la réflexion
des structures, des objets et des personnes.

Des interférences peuvent se produire à proximité des équipements portant le symbole
suivant :REMARQUE 1 : la plage de fréquence la plus élevée s’applique à 80 MHz et 800 MHz.
REMARQUE 2 : ces directives ne s'appliquent pas nécessairement dans tous les cas. La propagation électromagnétique est influencée par l'absorption et la réflexion
des structures, des objets et des personnes.
Les intensités de champ des émetteurs fixes, tels que les stations de base pour les téléphones radio (cellulaires/sans fil) et les radios mobiles terrestres, lesradios amateurs, les émissions de radio AM et FM et de télévision, ne peuvent théoriquement pas être prévues avec précision. Pour évaluer l'environnementélectromagnétique sous l'influence d'émetteurs RF fixes, une étude de site électromagnétique doit être envisagée. Si l'intensité du champ mesurée à l'emplacementd'utilisation de la lampe à polymériser dépasse le niveau de conformité RF applicable indiqué ci-dessus, vous devez vérifier le fonctionnement normal de la lampe à
polymériser VALO sans fil. Si des performances anormales sont observées, des mesures supplémentaires peuvent être nécessaires, telles que la réorientation ou ledéplacement de la lampe à polymériser.
b
Au-delà de la gamme de fréquences de 150 kHz à 80 MHz, les intensités de champ doivent être inférieures à 3 V/m.
a

30

31

VALO™ GRANDNLUITHARDINGSLAMP

Productomschrijving

Met zijn brede spectrum is de VALO™ Grand-polymerisatielamp ontworpen om alle lichthardende producten uit te harden in het golflengtebereik van 385-515 nm volgens ISO 10650. De
VALO Grand-polymerisatielamp maakt gebruik van de Ultradent VALO oplaadbare batterijen en batterijoplader. De polymerisatielamp past in een houder van een standaard behandelunitof kan in de meegeleverde houder op een zelf te kiezen plaats worden gemonteerd.

VALO Grand-productonderdelen:
•
•
•
•
•
•

Foto

Knoppen en lampjes

Opmerking:• Om kruisbesmetting te voorkomen en te voorkomen dat tandheelkundig composietmateriaal zich aan het lensoppervlak en de lampbehuizing hecht, moet er bij elk gebruik een
door Ultradent Products goedgekeurde beschermhuls over de VALO-polymerisatielamp worden gebruikt. De beschermhulzen zijn bedoeld voor gebruik bij één patiënt.

VALO Draadloos-lichtschild:

Het VALO Draadloos-lichtschild is ovaalvormig, kan voor maximaal gebruik worden gedraaid en kan worden gebruikt met een transparante barrièrehuls.

LegendeAan-/uitknop(pen)(Boven of onder)
Timinglampen
Batterij bijna leeg – indicator
Modus-/statuslampje
Knop voor wijzigen tijd/
modus(Onder)(Boven)Lees aandachtig alle instructies en het veiligheidsinformatieblad van de beschreven producten vóór gebruik.

Indicaties voor gebruik/beoogde doel

De lichtbron voor het uitharden van foto-geactiveerde dentale restauratiematerialen en kleefstoffen.

Warnings and Precautions
Risicogroep 2
LET OP Dit product straalt UV-straling uit. Blootstelling kan oog- of huidirritatie veroorzaken. Gebruik de geschikte afscherming.
LET OP Dit product kan gevaarlijke optische straling uitstralen. Kijk niet naar een werkende lamp. Dit kan schadelijk zijn voor de ogen.
• Kijk NIET rechtstreeks in de lichtbron. Patiënten, artsen en assistenten moeten altijd oranjekleurige oogbescherming tegen uv-licht dragen wanneer de polymerisatielamp in gebruikis.
• Om het risico van elektrische schokken te voorkomen, is het niet toegestaan wijzigingen aan deze apparatuur aan te brengen. Gebruik alleen de meegeleverde VALO-netvoeding ensnoeradapters van Ultradent Products. Als deze onderdelen beschadigd zijn, gebruik ze dan niet en bel de klantenservice van Ultradent Products om een vervanging te bestellen.
• Draagbare RF-communicatieapparatuur kan de prestaties verminderen als deze dichterbij dan 30 cm (12 in.) wordt gebruikt.
• Gebruik alleen goedgekeurde accessoires, kabels en voedingen om onjuiste bediening, verhoogde elektromagnetische emissies of verminderde elektromagnetische immuniteit tevoorkomen (zie het gedeelte Elektromagnetische emissies).
• Om het risico van elektrische brand bij gebruik van batterijen te vermijden:
⸰ Gebruik oplaadbare batterijen NIET samen met niet-oplaadbare batterijen of andere soorten batterijen.
⸰ Probeer NIET om niet-oplaadbare batterijen op te laden. Het opladen van niet-oplaadbare batterijen kan leiden tot letsel of materiële schade.
⸰ Batterijen, batterijcontacten, oplader of netvoeding NIET autoclaveren of besproeien met vloeistof van welke soort dan ook. Bij corrosie op de contacten van de batterijoplader:bel de klantenservice van Ultradent Products om een vervangend exemplaar te bestellen.
⸰ Laad batterijen NIET op in de buurt van ontvlambare materialen.
⸰ Bewaar de oplader NIET in de behandelkamer.
• Om het risico op letsel te voorkomen, mag u GEEN batterijen gebruiken die zijn gecorrodeerd (roest), gedeukt, een geur of vloeistoffen afgeven, een gescheurde of ontbrekende
verpakking hebben of op een andere manier beschadigd zijn. Bel de klantenservice van Ultradent Products om vervangende batterijen te bestellen.
• Gebruik alleen aanbevolen batterijen. Andere batterijen kunnen storingen veroorzaken.
• Om het risico van thermische irritatie of letsel te voorkomen, moet u uithardingscycli achter elkaar vermijden en orale weke delen van dichtbij niet langer dan 10 seconden
blootstellen in welke stand dan ook. Als langer uitharden nodig is, gebruik dan meerdere uithardingscycli met rustperioden tussen de cycli of gebruik een duaal uithardend productom verhitting van weke delen te voorkomen.
• Wees voorzichtig bij de behandeling van patiënten die last hebben van ongewenste fotobiologische reacties of gevoeligheden, patiënten die chemotherapie ondergaan of patiënten
die worden behandeld met fotosensibiliserende medicatie.
• Dit apparaat kan gevoelig zijn voor sterke magnetische of statische elektrische velden die de programmering kunnen verstoren. Als u vermoedt dat dit is gebeurd, haal dan even destekker uit het stopcontact en steek daarna de stekker weer in het stopcontact.
• Veeg de polymerisatielamp NIET af met bijtende of schurende reinigingsmiddelen, autoclaveer hem NIET en dompel hem NIET onder in een ultrasoon bad, ontsmettingsmiddel,
32

Stapsgewijze instructies
Voorbereiding

1) Laad de batterijen op voordat u de polymerisatielamp gebruikt (Zie gedeelte Onderhoud van batterijen).
2) Plaats voor het beste resultaat vóór elk gebruik een nieuwe beschermhuls over de polymerisatielamp, zodat er zo min mogelijk plooien voor de lens zitten.

1 - VALO Grand-polymerisatielamp
4 - Ultradent VALO oplaadbare batterijen
1 - Ultradent VALO batterijoplader met 12 V DC AC voedingsadapter, geschikt voor medische apparatuur
1 - proefverpakking VALO-beschermhulzen
1 - VALO Grand-veiligheidsschermpje
1 - Handstukhouder voor polymerisatielamp met dubbelzijdig plakband

Overzicht van bedieningen:reinigingsoplossing of vloeistof. Als de meegeleverde gebruiksinstructies niet worden opgevolgd, kan de polymerisatielamp onbruikbaar worden.
• Steek GEEN vingers, instrumenten of andere voorwerpen in het batterijcompartiment van de polymerisatielamp om beschadiging van de apparatuur te voorkomen.
• Om schade aan de apparatuur te voorkomen, mag u NIET proberen de gouden contacten of enig onderdeel van het batterijcompartiment schoon te maken. Bel de klantenservice van
Ultradent Products als er een reden tot bezorgdheid is.
• Om het risico op kruisbesmetting te voorkomen, zijn er beschermhulzen voor eenmalig gebruik.
• Om het risico op corrosie te verminderen, verwijdert u de beschermhuls na gebruik.
• Gebruik de polymerisatielamp niet als de lens beschadigd is om het risico van niet goed uitgehard restauratiemateriaal te verkleinen.

Gebruik

1) Elke stand wordt gebruikt voor het uitharden van tandheelkundige materialen met foto-initiatoren. Zie de Verkorte instructie standen voor aanbevolen uithardingstijden.
OPMERKING: De polymerisatielamp is zo geprogrammeerd dat hij eerst in de Standard Power-stand gaat en daarna naar achtereenvolgens de High Power Plus-stand en de Xtra
Power-stand. Om bijvoorbeeld van de Standard Power-stand naar de Xtra Power-stand te gaan, moet u eerst naar de High Power Plus-stand gaan en vervolgens naar de Xtra Powerstand.
2) De polymerisatielamp slaat het laatst gebruikte tijdsinterval en de laatste stand op, en zal hier standaard naar terugkeren wanneer er geschakeld wordt tussen de standen ofwanneer de batterijen verwijderd worden.

Werking
UITHARDINGSMODUS: Standaard vermogensmodus

TIMINGINTERVALLEN: 5, 10, 15, en 20 seconden.
• De uithardingslamp gaat standaard naar deze modus als deze AANVANKELIJK is ingeschakeld. Het modus-/statuslampje is groen en de vier groene timingslichten branden om destandaardvoedingsmodus aan te geven.
• Om de tijdsintervallen snel te wijzigen, drukt u op de tijd-/modusknop.
• Druk op de AAN/UIT-knop om het uitharden te starten. Als u wilt stoppen met uitharden voordat u een timinginterval hebt voltooid, drukt u nogmaals op de aan-/uitknop.

UITHARDINGSMODUS: Modus Hoog vermogen plus

TIMINGINTERVALLEN: 1, 2, 3, en 4 seconden.
• Druk, in de standaardvoedingsmodus op de knop voor het wijzigen van tijd/modus, houd deze 2 seconden ingedrukt en laat de knop daarna los. Het modus-/statuslampje is oranje
en de vier groene timing-lampjes gaan branden en knipperen, wat aangeeft dat de modus Hoog vermogen is ingeschakeld.
• Om de tijdsintervallen snel te wijzigen, drukt u op de tijd-/modusknop.
• Druk op een aan-/uitknop om te beginnen met uitharden. Om het uitharden te stoppen voordat een tijdsinterval is afgelopen, drukt u opnieuw op de aan-/uitknop.
• Om terug te keren naar de standaardvoedingsmodus houdt u de knop voor het wijzigen van tijd/modus 2 seconden ingedrukt en laat u die daarna weer los om naar de Extravermogen-modus te gaan. Houd opnieuw gedurende 2 seconden ingedrukt en laat los. Het modus-/statuslampje is groen en de vier groene timing-lampjes branden om destandaardmodus aan te geven.

UITHARDINGSMODUS: Modus Xtra vermogen

TIJDSINTERVALLEN: alleen 3 seconden (Opmerking.: De Extra vermogen-modus heeft een veiligheidsvertraging van 2 seconden aan het einde van elke uithardingscyclus om het
opwarmen tijdens opeenvolgende uitharding te beperken. Aan het einde van de vertraging geeft een pieptoon aan dat het apparaat weer kan worden gebruikt).
• Vanuit de standaardvoedingsmodus drukt u gedurende 2 seconden op de knop voor het wijzigen van de tijd/modus, laat u deze los, houdt hem opnieuw gedurende 2 seconden
ingedrukt en laat hem opnieuw los. Het modus-/statuslampje is oranje en knippert en drie van de groene tijdlampjes gaan branden en knipperen, waarmee de Xtra vermogenmodus wordt aangegeven.
• Druk op de AAN/UIT-knop om het uitharden te starten. Om het uitharden te stoppen voordat een tijdsinterval is afgelopen, drukt u opnieuw op de aan-/uitknop.
• Om terug te keren naar de standaardvoedingsmodus, houdt u de tijd/modus-knop gedurende 2 seconden ingedrukt en laat u deze weer los. Het modus-/statuslampje is groen en
de groene timing-lampjes branden om de standaardvoedingsmodus aan te geven.

Slaapstand: De uithardingslamp gaat na 60 seconden inactiviteit in de slaapstand, zoals wordt aangegeven door het langzaam knipperen van het modus-/statuslampje. Als u
het apparaat optilt of aanraakt, wordt de uithardingslamp geactiveerd en keert deze automatisch terug naar de laatst gebruikte instelling. Om de levensduur van de batterij temaximaliseren, laat u de uithardingslamp ongemoeid als u deze niet gebruikt.

Opruimen

1) Gooi de gebruikte beschermhulzen na elke patiënt weg bij het normale afval.
2) Zie het gedeelte Verwerking.

Instructies voor montage van de houder

1) De houder moet worden gemonteerd op een vlakke, olievrije ondergrond.
2) Reinig het oppervlak met alcohol.
3) Trek het plakband van de houder los.
4) Plaats de houder zo dat u de polymerisatielamp er naar boven toe uit moet halen. Druk de houder stevig aan.

Handleiding voor snelle modus:

33

Standaardvermogen
1000 mW/cm2

Modus

Modus Hoog vermogen
1600 mW/cm2

Extra vermogen
3200 mW/cm2

Aan-/uitknop
Modus/timing-leds

Tijdknoppen

Tijdopties

5s

10s

15s

20s

1s

2s

3s

4s

Alleen 3s

Om de tijd te wijzigen

Druk snel op de knop Tijd en laat deze onmiddellijk los om de tijdopties te doorlopen.

Om de modus te wijzigen

Houd knop Tijd gedurende 2 seconden ingedrukt en laat deze daarna los. De uithardingslamp gaat naar de volgende modus.

LegendeVaste LED’s

Knipperende LED’s

Handleiding voor snelle uitharding:
UithardingsstandVermogen* (mW)

Stralingssterkte* (mW/cm2)

Totale belichtingstijd (seconden)

Energie** (Joule)Standard

970

900

5/10/15/20

4,8J/9,7J/14,5J/19,4J

High Power Plus

1.620

1.500

1/2/3/4

1,6J/3,2J/4,8J/6,5J

Xtra

2.260

2.100

3

6,8J

*Nominaal vermogen op een afstand van 2 mm van de punt van de lens tot het bovenste oppervlak van de composiet.
**De waarden in het gedeelte over de totale energie per cyclus (joules) zijn naar boven afgerond op de dichtstbijzijnde tiende.

Snelle waarschuwingsgids:
Waarschuwing vermogensniveau

Waarschuwing temperatuur

Waarschuwing onderhoud

Waarschuwing LED

Vervang de batterijen

Laat afkoelen

Bel de klantendienst voor reparatie

Bel de klantendienst voor reparatie• Geen geluid• Knippert elke 2 sec.
• Kan worden gebruikt• Continu 3 pieptonen• Enkele snelle flits• Kan niet worden gebruikt• Batterij bijna leeg: langzaam
knipperend• Uitgeschakeld: 3 pieptonen,
knipperend• Kan niet worden gebruikt• 3 pieptonen• Knippert langzaam• Kan niet worden gebruikt

instrument om vastzittende composiet voorzichtig te verwijderen. Gebruik geen gereedschap waardoor de lens kan beschadigen.
2) Lichtmeters verschillen enorm van elkaar en zijn ontworpen voor specifieke lichtgeleidertips en lenzen. Ultradent Products raadt aan om regelmatig het vermogen in de stand Standard
Power te controleren. OPMERKING: het werkelijke numerieke vermogen zal vertekend zijn vanwege de onnauwkeurigheid van gewone lichtmeters en het aangepaste led-pakket in depolymerisatielamp
Onderhoud van batterijen
Batterijen opladen en vervangen
De polymerisatielamp wordt geleverd met 4 oplaadbare lithium-ijzerfosfaatbatterijen. Ultradent Products raadt aan om alleen VALO oplaadbare batterijen te gebruiken.
Batterijen opladen:
1) Zet de oplader en batterijen op een plaats waar ze niet in contact kunnen komen met vloeistoffen of ontsmettingsspray.
2) Plaats de batterijen in de oplader met het positieve (+) uiteinde in de richting van de indicatielampjes op de oplader.
3) Groene lampjes geven aan dat de batterijen klaar zijn voor gebruik.
4) Het duurt 1-3 uur om de batterijen op te laden. Laat de batterijen in de oplader zitten tot ze klaar zijn voor gebruik.
OPMERKING: als het rode lampje op de oplader niet groen wordt wanneer de batterijen langer dan drie uur zijn opgeladen, is de batterij mogelijk beschadigd en kan deze niet worden
opgeladen. Probeer een nieuwe batterij of bel de klantenservice van Ultradent Products om een nieuwe set oplaadbare batterijen te bestellen.
Batterijen vervangen/plaatsen:
1) Verwijder het kapje aan de onderkant door deze een kwartslag linksom te draaien.
2) Verwijder de batterijen.
3) Plaats nieuwe batterijen met de positieve (+) zijde eerst.
4) Bevestig het kapje aan de onderkant weer door het uit te lijnen en er zachtjes te duwen terwijl u met de klok mee draait. Het kapje klikt vast wanneer het goed bevestigd is.
5) Het apparaat is klaar voor gebruik.
Batterijen bijna leeg: de polymerisatielamp waarschuwt de gebruiker dat het tijd is om de batterijen te vervangen wanneer het indicatielampje voor lege batterijen rood knippert. Als debatterijlading te laag wordt, hoort u een waarschuwingssignaal met 3 pieptonen en kan de polymerisatielamp niet meer gebruikt worden totdat de batterijen opgeladen zijn of er nieuwe
batterijen zijn geplaatst. (Zie Verkorte instructie waarschuwingen)
Oplaadtijd en levensduur van de batterij: de levensduur van een volledig opgeladen batterij in de polymerisatielamp is afhankelijk van stand/tijdsinterval, batterijtype, gebruiksintensiteit
en led-efficiëntie.
• Aanbevolen oplaadinterval: laad wanneer nodig of wanneer de batterij-indicator oplicht. De levensduur hangt af van het gebruik.
• Extra batterijen: de polymerisatielamp wordt geleverd met 4 oplaadbare batterijen.
Reparatie door de fabrikant
1) Reparaties mogen alleen worden uitgevoerd door bevoegd onderhoudspersoneel. Ultradent Products voorziet het onderhoudspersoneel van documentatie om reparaties uit tevoeren.
2) Verwijder altijd de batterijen uit de polymerisatielamp en oplader als u apparaten opstuurt voor reparatie, service of kalibratie. Verpak batterijen, oplader, adapter enpolymerisatielamp apart in de retourdoos.
3) Verzend batterijen in overeenstemming met de plaatselijke voorschriften.
Garantie
Ultradent Products, Inc. ("Ultradent Products") garandeert dat dit product gedurende een periode van 5 jaar vanaf de aankoopdatum, mits het wordt gebruikt volgens debedieningsinstructies die bij het product zijn geleverd, (i) in alle materiële opzichten zal voldoen aan de specificaties die worden vermeld in de documentatie van Ultradent Products die
bij het product is geleverd; en (ii) vrij zal zijn van materiaal- en fabricagefouten.
Deze beperkte garantie is niet overdraagbaar en geldt alleen voor de oorspronkelijke koper en niet voor volgende eigenaars van het product. Deze beperkte garantie geldt niet voor
andere accessoires zoals, maar niet beperkt tot, batterijen, opladers, adapters of speciale lenzen. Deze beperkte garantie vervalt als het product defect raakt of beschadigd raakt als
gevolg van nalatigheid, misbruik, verkeerd gebruik, een ongeluk, aanpassingen, gepruts, wijzigingen of het niet opvolgen van de van toepassing zijnde gebruiksinstructies. Bijvoorbeeld:
een product dat is gevallen en beschadigd, valt niet onder deze garantie. Om in aanmerking te komen voor deze beperkte garantie, moet het aankoopbewijs (bijv. aankoopbon ofsoortgelijke documentatie) samen met het defecte product naar Ultradent Products worden gestuurd.
Een defect product dat voldoet aan de hierin uiteengezette garantievoorwaarden zal naar eigen goeddunken van Ultradent Products worden gerepareerd of vervangen. In geen geval
zal de aansprakelijkheid van Ultradent Products voor het product hoger zijn dan de door de koper betaalde aankoopprijs. In geen geval zal Ultradent Products aansprakelijk zijn voor
indirecte, incidentele, voorziene, onvoorziene, speciale of gevolgschade die voortvloeit uit of in verband staat met het gebruik van dit product.

Werkwijze

Dit product is bedoeld voor gebruik met een beschermhuls. Zie hierboven voor onderhoud en algemene reiniging

Opslag en verwijdering

Als u de uithardingslamp voor periodes langer dan 2 weken opslaat of als u deze voor transport wilt verpakken, verwijder dan altijd de batterijen. Als de batterijen lange tijd in het
apparaat blijven zitten zonder deze weer op te laden, is het mogelijk dat deze niet meer werken of niet meer opgeladen kunnen worden. Bewaar batterijen niet bij temperaturen boven
40 °C (104 °F) of in direct zonlicht.

OnderhoudOpslag en transport van de uithardingslamp:• Temperatuur: +10 °C tot +40 °C (+50 °F tot +104 °F)• Relatieve vochtigheid: 10% tot 95%• Omgevingsdruk: 500 hPa tot 1060 hPa

TOEGESTANE SCHOONMAAKMIDDELEN:• 70% isopropylalcohol• 70% ethanol

Technische overwegingen
Accessoires

Algemene reiniging van de polymerisatielamp
Bevochtig na elk gebruik een gaasje of zachte doek met een goedgekeurd desinfectiemiddel en veeg het oppervlak en de lens schoon. Door niet-toegestane schoonmaakmiddelen kan depolymerisatielamp beschadigd raken.

Algemene reiniging van het veiligheidsschermpje:
Reinig het VALO Cordless-veiligheidsschermpje met een willekeurig desinfecterend middel voor oppervlakken. NIET autoclaveren.
Reparatie
Door de gebruiker uitgevoerd onderhoud1) Gebruik een beschermhuls om te voorkomen dat tandheelkundig composiet zich aan het oppervlak van de lens hecht. Gebruik zo nodig een plastic of roestvrijstalen tandheelkundig
34

Houd u aan de plaatselijke richtlijnen voor afval en recycling voor het weggooien van elektronisch afval (zoals uithardingslampen, laders, batterijen en voedingen).

35

Item

CE-informatieTechnical information/Data

Effective Composite Curing Wavelength Bands:

VALO Draadloos lichtschild
SPECTRALE STRALING VERMOGEN (mW/nm)

EFFECTIEVE COMPOSIET UITHARDINGSGOLFLENGTEBANDEN(Bruikbaar golflengtebereik: 385-515 nm)

VALO lader

VALO batterijen

Effectief lager
golflengteband395-415 nm

Effectieve bovenste
golflengteband
440-480 nm

XtraHigh PowerStandard

WAVELENGTH (nm)Attribuut

Informatie/specificatie

Lens

Diameter 11,7 mm

Golfl. bereik• Bruikbaar golflengtebereik: 385 - 515 nm• Piekgolflengten: 395 - 415 nm en 440 - 480 nm
Nominale stralende rendementsvergelijkingsgrafiek

Meetinstrument

Lichtint. tab.

36

‡ Gigahertz-spectrumanalysator
Uitvoer

Totaal vermogen

Meteropening

15 mm

15 mm

Stand. Verm.
(±10%)

900 mW/cm²

970 mW

Modus Hoog
vermogen (±10%)1500 mW/cm²

1615 mW

Extra vermogen(±10%)

2100 mW/cm²

Stralingsemittantie varieert afhankelijk van
de mogelijkheden van het instrument, demeetmethode en de plaatsing van de lamp.
‡ Stralingsemittantie voldoet aan ISO 10650
indien gemeten met een Gigahertz-Optikspectrumanalysator.

2260 mW

VALO Grand uith.
lampClassificaties: IEC 60601-1 (veiligheid), IEC 606011-2 (EMC)

Gewicht:• Met batterijen: 190 g (6 oz.)
• Zonder batterijen: 150 g (5 oz.)
Afmetingen: 203 x 32,5 x 27 mm (8 x 1,28 x 1,06 in)Voed. lader

Uitgang - 12VDC bij 500mA
Ingang - 100VAC tot 240VAC
Ultradent P / N 5930 VALO ladervoeding met
internationale stekkerinzetstukken

Classificaties: IEC 60601-1 (veiligheid)
Snoerlengte - 1,8 m (6 ft)
De VALO-opladervoeding is een klasse II-voeding voor medische toepassing
en zorgt voor isolatie van het lichtnet

VALO lader

VALO 3,6VDC Lithium Iron Phosphate slimme batterijlader:• Automatisch uitschakelen indien volledig opgeladen• Automatische detectie van defecte batterijen• Beveiligingen: Thermisch, overbelasting, kortsluiting, omgekeerde polariteit⸰ Rode LED – Opladen⸰ Groene LED – Leeg of volledig opgeladen⸰ LED uit – Kortsluiting• Laadtijd: 1 – 3 uur
Classificatie: CE, WEEE
37

Overige informatie

Attribuut

Informatie/specificatie

VALO batt.

Oplaadbaar: Veilig chemische lithiumijzerfosfaat (LiFePO4) RCR123A• Werkspanning: 3,2 VDCClassificaties: UL, CE, RoHS, WEEE

Bedr.voorw.

Temperatuur: +10 °C tot +32 °C (+50 °F tot +90 °F)
Relatieve vochtigheid: 10% tot 95%
Omgevingsdruk: 700 hPa tot 1060 hPa

Bedrijfscycl.

De uithardingslamp is ontworpen voor gebruik op korte termijn. Bij maximale omgevingstemperatuur (32 °C) 1 minuut bij
opeenvolgend uit- en inschakelen, 30 minuten UIT (afkoeltijd).

Problemen oplossen
Als het probleem met de onderstaande suggesties niet kan worden opgelost, bel dan naar Ultradent op 800.552.5512. Bel buiten de VS uw Ultradent-distributeur of
tandheelkundig leverancier.
Probleem

Lamp wil niet branden

Lamp blijft niet branden tijdens degewenste tijd

De lamp hardt de harsen niet goed uit

Mogelijke oplossingen
1) Druk op de knop tijd/modus of aan/uit om te ontwaken uit de Power Save-modus.
2) Controleer of het rode indicatorlampje Batterij bijna leeg brandt.
3) Controleer of nieuwe batterijen op de juiste manier in de eenheid zijn geplaatst.
4) Als de rode en gele waarschuwings-LED's knipperen, betekent dit dat de uithardingslamp de interne
temperatuurveiligheidslimiet heeft bereikt. Laat de uithardingslamp 10 minuten afkoelen of gebruik een koele, vochtige
handdoek om het apparaat snel af te koelen.
5) Als de rode waarschuwing-LED knippert en de pieptonen aanhouden, bel dan de klantendienst van Ultradent voor
reparatie.
1) Controleer de lampjes voor modus en timing op de correcte tijdsinvoer.
2) Controleer of het indicatorlampje Batterij bijna leeg brandt.
3) Controleer of nieuwe batterijen op de juiste manier in de eenheid zijn geplaatst.
1) Controleer lens op uitgeharde hars-/composietresten.
2) Verifieer of de ledlampen werken met behulp van een goedwerkende amberkleurige oogbescherming.
3) Controleer het vermogen met een lichtmeter. Als u een lichtmeter gebruikt, adviseert Ultradent de uithardingslamp tecontroleren in de standaardvoedingsmodus.
OPMERKING: De echte numerieke uitgang zal gebogen zijn als gevolg van de onnauwkeurigheid van gewone lichtmeters en
het aangepaste LED-pakket dat door de uithardingslamp wordt gebruikt. Lichtmeters verschillen aanzienlijk en zijn ontworpenvoor specifieke lichtgeleidepunten en lenzen.
4) Controleer de vervaldatum op het uithardingshars.
5) Zorg ervoor dat de juiste techniek wordt gevolgd overeenkomstig de aanbevelingen van de fabrikant.

Batterijen worden niet opgeladen

1) Zorg ervoor dat de batterijen in de juiste richting in de lader worden geplaatst en laat de batterijen gedurende 1-3 uur
opladen.
2) Als de rode lampjes op de lader niet groen worden, bel dan de klantendienst van Ultradent om nieuwe batterijen en/of lader te bestellen.
3) Als er geen groene of rode lampjes op de lader zichtbaar zijn, bel dan de klantendienst van Ultradent om de lader en/of
wisselstroomadapter te bestellen of te vervangen.

De lader laadt de batterijen niet op1) Controleer of de stekker van de lader in de wisselstroomadapter is gestoken en de wisselstroomadapter in een goedwerkend stopcontact is gestoken.
2) Als de groene of rode lampjes op de lader niet zichtbaar zijn, bel dan de klantendienst van Ultradent voor een nieuwelader en/of wisselstroomadapter.

Kan modus of tijdsintervallen niet
veranderen

1) Houd de knop tijd/modus en de aan-/uitknop ingedrukt tot een reeks pieptonen aangeeft dat de uithardingslampje is
ontgrendeld.

Richtlijn en verklaring van de fabrikant betreffende voor elektromagnetische emissies
De uithardingslamp is bedoeld voor gebruik in de hieronder gespecificeerde elektromagnetische omgeving. De klant of gebruiker dient ervoor te zorgen dat het
systeem in een dergelijke omgeving wordt gebruikt.
Emissions Test

Conformiteit

Elektromagnetische omgeving – richtlijnen

RF-emissies
CISPR 11

Groep 1

De uithardingslamp maakt gebruik van batterijen en wordt niet beïnvloed door EMI-, RF- ofpiekstroombeveiliging.

RF-emissie
CISPR 11

Klasse B

Harmonische emissies
IEC 61000-3-2

NVT

De uithardingslamp gebruikt alleen elektrische en elektromagnetische energie voor de internefuncties. Daarom zijn eventuele RF-emissies zeer laag en zullen ze waarschijnlijk geen storing
veroorzaken in de nabijgelegen elektronische apparatuur.

Spanningsschommelingen/flikkeremissies IEC 61000-3-

NVT

Harmonische emissie- en spanningsschommelingstesten zijn niet van toepassing op deuithardingslamp omdat deze op batterijen werkt.
De uithardingslamp is geschikt voor gebruik op alle plaatsen, inclusief huishoudelijke
voorzieningen en gebouwen die direct zijn aangesloten op het openbare laagspanningsnetwerk
voor huishoudelijk gebruik.

Richtlijnen en verklaring van de fabrikant betreffende elektromagnetische immuniteit
De uithardingslamp is bedoeld voor gebruik in de hieronder gespecificeerde elektromagnetische omgeving. De klant of gebruiker dient ervoor te zorgen dat het
systeem in een dergelijke omgeving wordt gebruikt.
IMMUNITEIT-test

Testniveau IEC 60601

Conformiteitsniveau

Richtlijnen elektromagnetische omgeving

Elektrostatische ontlading (ESD)± 8 kV contact± 15 kV lucht± 8 kV contact± 15 kV lucht

De fysieke omgeving moet beperkt zijn tot het
volgende:
1) IP-Code: IP20
2) Niet onderdompelen in vloeistof.
3) Niet gebruiken in de buurt van brandbaar gas.
Eenheid is niet-APG en niet-AP.
4) Vochtigheidsbereik voor opslag: 10% - 95%
5) Temperatuurbereik voor opslag: 10 °C - 40 °C

Snelle elektrische transiënten/
burst
IEC 61000-4-4± 2 kV voor
stroomtoevoerleidingen± 1 kV voor invoer-/uitvoerleidingen

Overspanning
IEC 61000-4-5± 1 kV leiding naar leiding± 2 kV leiding naar de aarde

Spanning, spanningsdip,
kortsluiting en onderbreking en
schommeling van de ingangenvan de netvoeding<5% U(>95% daling van U gedurende0,5 cyclus)± 2 kV voor
De uithardingslamp werkt op batterijen en is niet
stroomtoevoerleidingengeschikt voor aansluiting op netvoeding.
Opmerking 1: De uithardingslamp
heeft geen I/0-poorten
Omdat de uithardingslamp op batterijen werkt, isdeze niet onderhevig aan elektrische overspanningen,spanningspieken, spanningsdalingen, korte± 1 kV leiding naar leiding
onderbrekingen, onderbrekingen of schommelingen± 2 kV leiding naar de aarde
van de netvoeding.
5% U
De bijgeleverde batterijlader kan onderhevig zijn aan(>95% daling van U gedurende
het bovenstaande, maar is los van en niet kritisch voor
0,5 cyclus)
de werking van de VALO Draadloos uithardingslamp.
40% U
Als de batterijspanning van de uithardingslamp daalt(60% daling van U gedurende
naar 4 VDC, zal het apparaat niet meer werken. De
5 cycli)
VALO Draadloos uithardingslamp zal uitschakelen.
Wanneer nieuwe batterijen worden geplaatst en de
70% U
juiste vermogensniveaus worden hersteld, zal de(30% daling van U gedurende
uithardingslamp opnieuw opstarten en terugkeren
25 cycli)naar de staat waarin deze stond voor het uitvallen vande voeding. De uithardingslamp zal zichzelf herstellen<5% U
als de voeding is weggevallen.
(>95% daling van U gedurende
5 seconden)
Opmerking 2: Herstelt zelf de

IEC 61000-4-2

IEC 61000-4-11

40% U(60% daling van U gedurende
5 cycli)
70% U(30% daling van U gedurende
25 cycli)<5% U(>95% daling van U gedurende
5 seconden)

Vermogensfr. (50/60 Hz)
magn. Veld

30 A/m30 A/mIEC 61000-4-8

De magnetische velden van de voeding moetenzich op niveaus bevinden die kenmerkend zijn voor
een typische locatie in een typische woon-, zorg-,
commerciële, ziekenhuis- of militaire omgeving.

Opmerking 1: De uithardingslamp is niet uitgerust met of toegankelijke I/O-lijnen.
3839Richtlijn en verklaring van de fabrikant betreffende elektromagnetische immuniteit voor niet-live ondersteunende systemen
De uithardingslamp is bedoeld voor gebruik in de hieronder gespecificeerde elektromagnetische omgeving. De klant of gebruiker dient ervoor te zorgen dat het
systeem in een dergelijke omgeving wordt gebruikt.
IMM.-test

Geleiding RF

IEC 61000-4-6

Testniveau IEC 60601

3 Vrms150 kHz tot 80 MHz

Conformiteitsniveau Richtlijnen elektromagnetische omgeving

3 VrmsDraagbare en mobiele RF-communicatieapparatuur mag niet dichter bij een onderdeel
van de uithardingslamp, inclusief kabels, worden gebruikt dan de aanbevolen
scheidingsafstand, berekend op basis van de vergelijking die van toepassing is op defrequentie van de zender.
Aanbevolen scheidingsafstand:

150 kHz tot 80 MHz
80 MHz tot 800 MHz

Uitgestraalde RF

3 V/m

3 V/m

800 MHz tot 2,5 GHz
P is het maximale uitgangsvermogen van de zender in watt (W) volgens de specificaties vande fabrikant van de zender, en d de aanbevolen scheidingsafstand in meter (m).
De veldsterkte van vaste RF-zenders, zoals bepaald door een elektromagnetische inspectie
ter plaatsea, moet lager zijn dan het conformiteitsniveau in elk frequentiebereikb.

IEC 61000-4-3

80 MHz tot 2,5 GHz

80 MHz tot 2,5 GHz

Er kan storing optreden in de buurt van apparatuur die is gemarkeerd met het volgende
symbool:

OPMERKING 1 Het hogere frequentiebereik is van toepassing bij 80 MHz en 800 MHz.
OPMERKING 2 Deze richtlijnen zijn mogelijk niet in alle situaties van toepassing. Elektromagnetische overdracht wordt beïnvloed door absorptie en weerkaatsing van
structuren, objecten en personen.
De veldsterktes van vaste zenders, zoals basisstations voor radiotelefoons (mobiel/draadloos) en portofoons, amateurradio‘s, AM- en FM-radio-uitzendingen
en tv-uitzendingen kunnen theoretisch niet nauwkeurig worden voorspeld. Om de elektromagnetische omgeving te beoordelen met vaste RF-zenders, moet een
elektromagnetische onderzoek ter plaatse overwogen worden. Als de gemeten veldsterkte op de locatie waar het polymerisatielicht wordt gebruikt, hoger is dan
het toepasselijke RF-conformiteitsniveau dat hierboven wordt vermeld, moet de VALO Draadloos uithardingslamp worden geobserveerd om de normale werking
te controleren. Als abnormale prestaties worden waargenomen, kunnen aanvullende maatregelen nodig zijn, zoals het opnieuw richten of verplaatsen van deuithardingslamp.
bOver het frequentiebereik van 150 kHz tot 80 MHz moeten de veldsterktes minder zijn dan 3 V/m.

Richtlijnen en verklaring van de fabrikant betreffende aanbevolen scheidingsafstanden tussen draagbare en mobiele RF-communicatieapparatuur en de VALO
Draadloos uithardingslamp
De uithardingslamp is bedoeld voor gebruik in een elektromagnetische omgeving waarin uitgestraalde RF-storingen onder controle zijn. De gebruiker
van de uithardingslamp kan elektromagnetische storing helpen voorkomen door een minimumafstand aan te houden tussen draagbare en mobiele
RF-communicatieapparatuur (zenders) en de uithardingslamp zoals hieronder wordt aanbevolen, afhankelijk van het maximale uitgangsvermogen van decommunicatieapparatuur.
Scheidingsafstand volgens de frequentie van de zender(meter)Nominaal maximaal uitgangsvermogen
van zender(P in Watt)

150 kHz – 80 MHz

80 MHz – 800 MHz

800 MHz – 2,5 GHz

0,01

0,12 meter

0,035 meter

0,07 meter

0,1

0,37 meter

0,11 meter

0,22 meter

1

1,7 meter

0,35 meter

0,7 meter

10

3,7 meter

1,11 meter

2,22 meter

100

11,7 meter

3,5 meter

7,0 meter

De uithardingslamp is getest volgens IEC 60601-1-2: 2014 en onder velden met een stralingssterkte 10 V / m tussen 80 MHz tot 2,5 GHz doorgegeven. De waarde
van 3Vrms komt overeen met V1 en de waarde 10 V/m komt overeen met E1 in de bovenstaande formules.
Voor zenders met een maximaal uitgangsvermogen dat hierboven niet vermeld wordt, kan de aanbevolen scheidingsafstand d in meters (m) worden geschat
met behulp van de toepasselijke formule voor de frequentie van de zender, waarbij P het maximale uitgangsvermogen van de zender in watt (W) is, volgens despecificaties van de fabrikant van de zender.
OPMERKING 1 Bij 80 MHz en 800 MHz is de scheidingsafstand voor het hogere frequentiebereik van toepassing.
OPMERKING 2 Deze richtlijnen zijn mogelijk niet in alle situaties van toepassing. Elektromagnetische overdracht wordt beïnvloed door absorptie en weerkaatsing van
structuren, objecten en personen.

a

40

41

VALO™ GRANDIT

FOTOPOLIMERIZZATRICE
Descrizione del prodotto

Grazie al suo ampio spettro, la lampada fotopolimerizzante VALO™ grand è concepita per la polimerizzazione di tutti i materiali dentali fotoindurenti a lunghezza d'onda comprese tra i
385-515nm, conformemente allo standard ISO 10650. La lampada fotopolimerizzante VALO Grand deve essere utilizzata con il caricabatterie e le batterie ricaricabili Ultradent VALO. La
lampada fotopolimerizzante è progettata per essere riposta nella faretra del riunito standard o per essere montata su misura utilizzando il supporto compreso nel kit.

Componenti del prodotto Valo Grand:
•
•
•
•
•
•

1 – Lampada fotopolimerizzante VALO Grand4 – Batterie ricaricabili Ultradent VALO
1 – Caricabatterie Ultradent VALO con adattatore di alimentazione CA 12VDC di grado medico
1 – Confezione campione di guaine di protezione VALO
1 – Schermo protettivo VALO Grand1 – Staffa di montaggio per il manipolo dotata di nastro biadesivo

Panoramica dei comandi:

Foto

Procedure di utilizzo
Preparazione

1) Prieš naudodami kietinimo lempą, įkraukite akumuliatorius (žr. skyrių Akumuliatoriaus priežiūra).
2) Prieš kiekvieną naudojimą ant kietinimo lempos uždėkite naują apsauginę movą, kad lęšis kuo mažiau raukšlėtųsi ir būtų pasiekti geriausi rezultatai.
• Per evitare la contaminazione incrociata e che il materiale dentale composito aderisca alla superficie della lente e del corpo del manico, è necessario utilizzare una guaina protettiva
approvata da Ultradent sulla lampada fotopolimerizzante VALO a ogni utilizzo. Le guaine protettive sono destinate all'uso esclusivo di un singolo paziente.

Schermo protettivo VALO Cordless:

Lo schermo protettivo VALO Cordless ha forma ovale, può essere ruotato per massimizzarne l’utilizzo e può essere utilizzato con una guaina protettiva trasparente.

Pulsanti e spie

LegendaSpie tempo
Batteria in esaurimento - Spia
Spia programma/stato
Pulsante di cambio tempo/
programma(Parte superiore)

Per tutti i prodotti descritti, leggere attentamente e comprendere tutte le istruzioni e le informazioni sulla SDS prima dell'uso.

Indicazioni per l'uso/ Utilizzo previsto

Sorgente di illuminazione per la polimerizzazione degli adesivi e dei materiali da restauro dentali fotoattivabili.

Avvertenze e precauzioni
Gruppo di rischio 2
ATTENZIONE: Emissione di raggi UV da parte del prodotto. L'esposizione sugli occhi o sulla pelle possono causare irritazione. Utilizzare schermature appropriate.
ATTENZIONE: possibili radiazioni ottiche pericolose emesse da questo prodotto. Non fissare la lampada quando è in funzione. Può essere dannoso per gli occhi.
• NON guardare direttamente la sorgente luminosa. Il paziente, il medico e gli assistenti devono sempre indossare una protezione per gli occhi di colore ambrato quando la lampada
fotopolimerizzante è in uso.
• Per evitare il rischio di scosse elettriche, non sono consentite modifiche a questo apparecchio. Utilizzare solo l'alimentatore Ultradent VALO e gli adattatori inclusi. Se questi
componenti sono danneggiati, non utilizzarli e chiamare il Servizio Clienti Ultradent per ordinarne la sostituzione.
• I dispositivi di comunicazione in radiofrequenza (RF) portatili possono ridurre le prestazioni se utilizzate a una distanza inferiore a 30 cm (12 in).
• Utilizzare solo accessori, cavi e alimentatori autorizzati onde evitare un funzionamento improprio, un aumento delle emissioni elettromagnetiche o una riduzione della immunità
elettromagnetica (fare riferimento alla sezione Emissioni elettromagnetiche).
• Per evitare il rischio di incendio elettrico relativo alla manipolazione delle batterie:⸰ NON combinare batterie ricaricabili e batterie non ricaricabili o altri tipi di batteria.
⸰ NON provare a caricare batterie non ricaricabili. La ricarica di batterie non ricaricabili può provocare lesioni o danni materiali.
⸰ NON sterilizzare in autoclave o spruzzare batterie, contatti delle batterie, caricabatterie o alimentatore CA con liquidi di alcun tipo In caso di comparsa di segni di corrosione suicontatti del caricabatterie, chiamare il Servizio Clienti Ultradent per ordinarne la sostituzione.
⸰ NON caricare le batterie in prossimità di materiali infiammabili.
⸰ NON tenere il caricatore in sala operatoria.
• Per evitare il rischio di lesioni, NON utilizzare batterie corrose (ruggine), ammaccate, che emettono odori o liquidi, che hanno un involucro strappato o mancante o che sono
altrimenti danneggiate. Chiamare il Servizio Clienti Ultradent per ordinare batterie di ricambio.
• Usare soltanto le batterie raccomandate. Altre batterie possono causare malfunzionamenti.
• Per prevenire il rischio di irritazione o lesioni termiche, evitare cicli di polimerizzazione consecutivi e non esporre i tessuti molli orali a distanza ravvicinata per più di 10 secondi inalcuna modalità. Se sono necessari tempi di polimerizzazione più lunghi, effettuare più cicli di polimerizzazione con periodi di riposo tra un ciclo e l'altro, o usare un prodotto a doppia
polimerizzazione per evitare di riscaldare i tessuti molli.
• Usare con cautela durante il trattamento di pazienti che soffrono di reazioni fotobiologiche avverse o soggetti a sensibilità, di pazienti che sono sottoposti a trattamenti chemioterapici
o di pazienti in trattamento con farmaci fotosensibilizzanti.
• Questa unità può essere sensibile a forti campi elettrici magnetici o statici che potrebbero interferire con la programmazione. Se si sospetta che si sia verificato il caso sopra
descritto, scollegare momentaneamente l'unità e poi ricollegarla alla presa di corrente.
• NON pulire la lampada fotopolimerizzante con detergenti caustici o abrasivi, non mettere in autoclave o immergere in alcun tipo di bagno a ultrasuoni, soluzioni disinfettanti o42detergenti né in alcun tipo di liquido. La mancata osservanza delle istruzioni di trattamento incluse può rendere inutilizzabile la lampada fotopolimerizzante.
Per evitare di danneggiare l'apparecchiatura, NON inserire dita, strumenti o altri oggetti nel vano batterie della lampada polimerizzante.
Per evitare di danneggiare l’apparecchiatura, NON cercare di pulire i contatti dorati o qualsiasi parte del vano batterie. In caso di dubbi, contattare il Servizio Clienti Ultradent.
Per evitare il rischio di contaminazione crociata, le guaine protettive sono destinate a un utilizzo monouso.
Per ridurre il rischio di corrosione, rimuovere la guaina protettiva dopo l'uso.
Per ridurre il rischio di resine non sufficientemente polimerizzate, non utilizzare la lampada fotopolimerizzante se la lente è danneggiata.

Nota:

Pulsante di accensione(Parte superiore o inferiore)(Parte inferiore)

•
•
•
•
•

Uso

1) Ogni modalità di potenza viene utilizzata per la polimerizzazione dei materiali dentali con fotoiniziatori. Consultare la Guida rapida ai programmi per i tempi di polimerizzazione
consigliati.
NOTA: La lampada fotopolimerizzante è programmata per scorrere sequenzialmente dal programma Standard Power attraverso il programma High Power Plus al programma Xtra
Power. Ad esempio, per passare dal programma Standard Power fino al programma Xtra Power, è necessario passare dal programma High Power Plus.
2) La lampada fotopolimerizzatrice memorizza l’ultimo intervallo di tempo utilizzato in ogni programma e vi ritorna per impostazione predefinita quando si cambia il programma oquando si rimuovono le batterie.

Funzionamento
PROGRAMMA DI FOTOPOLIMERIZZAZIONE: Programma potenza Standard

INTERVALLI DI TEMPO: 5, 10, 15, e 20 secondi.
• Quando si accende la lampada fotopolimerizzatrice PER LA PRIMA VOLTA, la lampada passa per impostazione predefinita a questo programma. La spia “Programma/Stato” saràverde e le quattro spie “Tempo” verdi si illumineranno, indicando il programma potenza standard.
• Per modificare gli intervalli di tempo, premere rapidamente il pulsante “Tempo/Programma”.
• Premere il pulsante di “Accensione” per fotopolimerizzare. Per interrompere la fotopolimerizzazione prima del completamento di un intervallo di tempo, premere nuovamente ilpulsante “Accensione”.

PROGRAMMA DI FOTOPOLIMERIZZAZIONE: Programma potenza Alta Plus

INTERVALLI DI TEMPO: 1, 2, 3, e 4 secondi.
• Nel programma potenza Standard, premere e tenere premuto il pulsante di modifica Tempo/Programma per 2 secondi e rilasciarlo. La spia Programma/Stato sarà arancione e lequattro spie tempo verdi si illumineranno e lampeggeranno, indicando il programma Alta potenza.
• Per modificare gli intervalli di tempo, premere rapidamente il pulsante Tempo/Programma.
• Premere uno dei pulsanti di accensione per fotopolimerizzare. Per interrompere la fotopolimerizzazione prima del completamento di un intervallo di tempo, premere nuovamente
il pulsante di accensione.
• Per tornare al programma potenza Standard, premere e tenere premuto il pulsante di modifica Tempo/Programma per 2 secondi e rilasciarlo, per passare al programma potenza
Xtra. Premere e tenere premuto il pulsante nuovamente per 2 secondi e rilasciarlo. La spia Programma/Stato sarà verde e le quattro spie tempo verdi si illumineranno, indicando il
programma Standard.

PROGRAMMA DI FOTOPOLIMERIZZAZIONE: Programma potenza Xtra

INTERVALLO DI TEMPO: Solo 3 secondi (Nota: Il programma potenza Xtra ha un blocco di sicurezza di 2 secondi al termine di ogni ciclo di fotopolimerizzazione per limitare il
riscaldamento durante polimerizzazioni consecutive. Alla fine del tempo di blocco, un segnale acustico indica che è possibile usare nuovamente l’unità).
• Nel programma potenza Standard, premere il pulsante di modifica Tempo/Programma per 2 secondi, rilasciarlo, tenerlo premuto nuovamente per 2 secondi e rilasciarlo. La spia
Programma/Stato sarà arancione e lampeggerà, e tre delle spie tempo verdi si illumineranno e lampeggeranno, indicando il programma potenza Xtra.
• Premere il pulsante di accensione per fotopolimerizzare. Per interrompere la fotopolimerizzazione prima del completamento di un intervallo di tempo, premere nuovamente ilpulsante di accensione.
• Per tornare al programma potenza Standard, premere e tenere premuto il pulsante Tempo/Programma per 2 secondi e rilasciarlo. La spia Programma/Stato sarà verde e le spie
tempo verdi si illumineranno, indicando il programma potenza Standard.

Modalità “sospensione”: La lampada fotopolimerizzante entrerà in modalità “Sospensione” dopo 60 secondi di inattività, come indicato dalla spia programma/stato che lampeggerà
lentamente. Sollevare o toccare l'unità per riattivare la lampada fotopolimerizzante che tornerà automaticamente all'ultima impostazione utilizzata. Per massimizzare la durata dellabatteria, lasciare indisturbata la lampada fotopolimerizzante quando non è in uso.

Pulizia

1) Gettare via le guaine protettive usate nei rifiuti normali dopo ogni paziente.
2) Consultare la sezione Trattamento.

Istruzioni per la staffa di montaggio1) La staffa deve essere montata su una superficie piana, non oleosa.
2) Pulire la superficie con alcol isopropilico.
3) Staccare la pellicola dal nastro adesivo della staffa.
4) Posizionare la staffa in modo che la lampada fotopolimerizzante possa essere sfilata verso l’alto. Premere bene in posizione.

Guida rapida ai programmi:
43

Potenza Standard
1000 mW/cm2

Programma

Potenza Alta Plus
1600 mW/cm2

Potenza Xtra
3200 mW/cm2

Detergere lo schermo protettivo VALO Cordless utilizzando un qualsiasi disinfettante per superfici. NON sterilizzare in autoclave.
Riparazione
Manutenzione eseguibile dall'utente
1) Utilizzare una guaina protettiva per evitare che la resina dentale aderisca alla superficie della lente. Se necessario, usare uno strumento odontoiatrico non diamantato
per rimuovere con cautela la resina adesa. Non utilizzare strumenti che possano danneggiare la lente.
2) I radiometri differiscono notevolmente tra di loro e sono realizzati per specifiche lenti e puntali ottici. Ultradent raccomanda di controllare regolarmente la potenza
in uscita nel programma Standard Power. NOTA: il valore numerico della potenza può non corrispondere al valore effettivo a causa dell'inadeguatezza dei radiometricomuni e del pacchetto LED personalizzato nella lampada fotopolimerizzante.
Manutenzione della batteria
Carica e sostituzione delle batterie
La lampada polimerizzante è dotata di 4 batterie ricaricabili al litio-ferro-fosfato. Ultradent raccomanda di utilizzare esclusivamente batterie ricaricabili VALO.

Solo 3s

Come caricare le batterie:1) Collocare il caricabatterie e le batterie in un'area in cui non entrino in contatto con liquidi o spray disinfettanti.
2) Inserire le batterie nel caricabatterie con il polo positivo (+) rivolto verso le spie luminose del caricabatterie.
3) Le spie verdi indicano che le batterie sono pronte per l'uso.
4) La carica delle batterie richiede 1-3 ore. Lasciare le batterie nel caricabatterie fino al loro utilizzo.
NOTA: se la spia rossa del caricabatterie non diventa verde dopo aver messo in carica le batterie per più di 3 ore, ciò significa che queste potrebbero essere deteriorate(danneggiate) e pertanto non possono essere caricate. Provare a inserire una batteria nuova o contattare il Servizio Clienti Ultradent per ordinare un nuovo set dibatterie ricaricabili.

Pulsante di accensione
LED programma/tempo

Pulsante tempo

Opzioni tempo

5s

10s

15s

20s

1s

2s

3s

4s

Cambio tempo

Premere e rilasciare velocemente il pulsante tempo per scorrere tra le opzioni tempo.

Cambio programmi

Premere e tenere premuto il pulsante tempo per 2 secondi e rilasciarlo. La lampada fotopolimerizzante passerà al programma
successivo.

LegendaLED a luce fissa

LED lampeggianti

Guida rapida alla fotopolimerizzazione:
Programma dipolimerizzazione

Potenza* (mW)

Irraggiamento* (mW/cm2)

Tempo di esposizione totale (secondi)

Energia** (Joule)

Come sostituire/inserire le batterie:
1) Rimuovere il tappo di chiusura ruotandolo in senso antiorario per un quarto di giro.
2) Rimuovere le batterie.
3) Inserire le batterie cariche con il polo positivo (+) per primo.
4) Applicare nuovamente il tappo di chiusura, allineandolo e premendo delicatamente mentre lo si ruota in senso orario. Raggiunta la posizione corretta di chiusura si
avvertirà uno scatto.
5) L'unità è pronta per l'uso.

Standard

970

900

5/10/15/20

4.8J/9.7J/14.5J/19.4J

High Power Plus

1,620

1,500

1/2/3/4

1.6J/3.2J/4.8J/6.5J

Xtra

2,260

2,100

3

6.8J

Batterie in esaurimento: la lampada fotopolimerizzante segnala all'utente che è arrivato il momento di sostituire le batterie quando la spia di batteria in esaurimento
lampeggia in rosso. Se il livello di carica delle batterie è troppo basso, si avvertiranno tre segnali acustici e non sarà possibile eseguire ulteriori operazioni fino a quando
le batterie non saranno ricaricate o sostituite con batterie nuove. (Consultare la voce Guida rapida agli avvisi)
Tempo di carica e durata della batteria: la durata della batteria completamente carica nella lampada fotopolimerizzante dipende dal Programma, dall'intervallo di tempo,
dal tipo di batteria, dalla frequenza di utilizzo e dall'efficienza del LED.
• Intervallo di ricarica consigliato: ricaricare le batterie quando necessario o quando si accende la spia della batteria in esaurimento. L'aspettativa di durata della batteria
dipende dall'uso.
• Batterie supplementari: la lampada fotopolimerizzante presenta 4 batterie ricaricabili in dotazione.
Garanzia**Uscita nominale a una distanza di 2 mm dalla punta della lente alla superficie superiore del composito.
**I valori nella sezione energia totale per ciclo (joule) sono arrotondati al decimo superiore.

Guida rapida agli avvisi:
Avviso livello intensità

Avviso temperatura

Avviso di servizio

Avviso LED

Sostituire le batterie

Lasciare raffreddare

Chiamare il Servizio Clienti per lariparazione

Chiamare il Servizio Clienti per lariparazione• Batteria in esaurimento:
lentamente lampeggiante• Spegnimento: 3 segnali acustici,
lampeggiante• Funzionamento non consentito• 3 segnali acustici• Lentamente lampeggiante• Funzionamento non consentito• Nessun suono• Lampeggiante ogni 2 secondi• Funzionamento consentito• 3 segnali acustici continui• Lampeggiamento singolo rapido• Funzionamento non consentito

Ultradent Products, Inc. ("Ultradent") garantisce che questo prodotto, per un periodo di 5 anni dalla data di acquisto, se utilizzato secondo le istruzioni operative incluse nel prodotto, (i)è conforme in tutti gli aspetti materiali alle specifiche indicate nella documentazione di Ultradent che accompagna il prodotto; e (ii) è privo di difetti di materiale e di lavorazione.
La presente garanzia limitata non è trasferibile e si applica esclusivamente all'acquirente originale e non si estende ai successivi proprietari del prodotto. La presente garanzia limitata
non copre altri componenti accessori quali, a titolo esemplificativo ma non esaustivo, batterie, caricabatterie, adattatori o lenti adattive. La presente garanzia limitata è nulla se il
prodotto si guasta o si danneggia a causa di negligenza, abuso, uso improprio, incidente, modifica, manomissione, alterazione o mancata osservanza delle istruzioni d'uso applicabili. A
titolo puramente esemplificativo, un prodotto caduto e danneggiato non è coperto dalla presente garanzia. Per poter usufruire della presente garanzia limitata, è necessario presentare
a Ultradent la prova d'acquisto (ad es. scontrino fiscale o documentazione simile) insieme al prodotto difettoso.
Un prodotto difettoso che soddisfa le condizioni di garanzia qui esposte sarà, a esclusiva discrezione di Ultradent, riparato o sostituito. In nessun caso la responsabilità economica di
Ultradent per il prodotto supererà il prezzo d'acquisto pagato dall'acquirente. In nessun caso Ultradent sarà responsabile per danni indiretti, incidentali, previsti, imprevisti, speciali oconsequenziali derivanti da o in relazione all'uso di questo prodotto.

Trattamento

Questo prodotto è destinato a essere utilizzato con una guaina protettiva. Consultare la sezione di cui sopra per la manutenzione e la pulizia generale.

Conservazione e smaltimento

Manutenzione
Pulizia generale della lampada fotopolimerizzante
Dopo ogni utilizzo, inumidire una garza o un panno morbido con un disinfettante per superfici approvato e pulire la superficie e la lente. I detergenti non autorizzati
possono danneggiare la lampada fotopolimerizzante.
DETERGENTI AUTORIZZATI:• 70% di alcool isopropilico• 70% di etanolo
Pulizia generale dello schermo protettivo:
44

Se la lampada fotopolimerizzatrice viene conservata per periodi superiori a 2 settimane o imballata per il trasporto, rimuovere sempre le batterie. Se le batterie rimangono nell'unità perlunghi periodi di tempo senza essere ricaricate, potrebbero non funzionare più o non essere più ricaricabili. Non conservare le batterie a temperature superiori a 60 °C (104°F o alla luce
diretta del sole.
Stoccaggio e trasporto della lampada fotopolimerizzatrice:• Temperatura: da +10°C a +40°C• Umidità relativa: dal 10% al 95%• Pressione ambientale: da 500 hPa a 1060 hPa
Per lo smaltimento dei rifiuti elettronici (per lampade per la fotopolimerizzazione, caricatori, batterie e alimentatori), seguire le istruzioni locali in materia di rifiuti e riciclaggio.

Considerazioni tecniche
Accessori45

Guaine protettive VALO

Informazioni/Dati tecnici

Informazioni CE

MDSS GmbH
Schiffgraben 4130175 Hanover
Germania

Fabbricato da:
Prodotti TIDI, LLC.
Unità aziendale 570
Neenah, WI 54956
Fabbricato negli Usa

Bande di lunghezza d'onda efficaci per la fotopolimerizzazione del composito:
Distribuito da:
Ultradent Products Inc
505 West Ultradent Drive (10200
South)South Jordan, UT 84095
Stati Uniti d'America

BANDE DI LUNGHEZZE D'ONDA EFFICACI PER LA POLIMERIZZAZIONE DEICOMPOSITI(Gamma di lunghezze d'onda utilizzabili: 385-515 nm)
POTENZA RADIANTE SPETTRALE (mW/nm)

Articolo

Schermo protettivo VALO senza fili

Caricabatterie VALO

Batterie VALO

Inferiore effettivo
banda di lunghezza
d'onda395-415 nm

Banda di lunghezza
d'onda superiore effettiva
440-480 nm

XtraHigh PowerStandard

WAVELENGTH (nm)Attributo

Informazioni/Specifiche

Lente

Diametro 11,7 mm

Intervallo di
lunghezza d'onda• Intervallo di lunghezza d'onda utilizzabile: 385 - 515nm• Lunghezze d'onda di picco: 395 - 415 nm e 440 - 480 nm
Nominal Radiant Exitance Comparison Chart
Strumento dimisurazione

Tabelladell'Intensità
luminosa

46‡ Analizzatore a spettro Gigahertz
Emettenza

Intensità totale

Apertura del metro

15 mm

15 mm

Potenza Standard(±10%)

900 mW/cm²

970 mW

High Power Plus(±10%)

1500 mW/cm²

1615 mW

Xtra Power (±10%)2100 mW/cm²

L'emettenza radiale varia in base alla capacitàdello strumento, al metodo di misurazione e al
posizionamento della lampada.
‡ L’emettenza radiale è conforme allo standard
ISO 10650 se misurato con analizzatori aspettro ottico Gigahertz.

2260 mW

Lampada fotopolimerizzatrice VALOGrandClassificazione: IEC 60601-1 (sicurezza), IEC
60601-1-2 (EMC)

Peso:• Con batterie: 190 g (6 oz)• Senza batterie: 150 g (5 oz)
Dimensioni: (8 x 1,28 x 1,06) pollici, (203 x 32,5 x 27) mm

Alimentatore per
caricabatterie

Uscita - da 12VDC a 500mA
Ingresso - da 100VAC a 240VAC
Alimentatore per caricabatterie Ultradent P/N5930 VALO con adattatori internazionali

Classificazione: IEC 60601-1 (sicurezza)
Lunghezza del cavo - 1,8 m (6 ft)
L'alimentatore del caricabatterie VALO è un alimentatore di grado medicale- classe II e fornisce l'isolamento dalla rete di alimentazione

Carica-batterie
VALO

Caricabatterie VALO 3,6 VCC Smart al litio-ferro-fosfato:• Spegnimento automatico quando la batteria è completamente carica.
• Individuazione automatica delle batterie difettose• Protezioni: termica, sovraccarico, cortocircuito, inversione di polarità
⸰ LED rosso - Caricamento⸰ LED verde - Vuoto o completamente caricato
⸰ LED spento - Cortocircuito• Tempo di carica: 1 - 3 ore
Classificazione: CE, RAEE47

Informazioni varie

Attributo

Informazioni/Specifiche

Batterie VALO

Ricaricabili: RCR123A al litio-ferro-fosfato, chimica sicura (LiFePO4)• Tensione di funzionamento: 3,2 VCCClassificazione: UL, CE, RoHS, RAEE

Condizioni
operative

Temperatura: da +10°C a +32°C (da +50°F a +90°F)
Umidità relativa: dal 10% al 95%
Pressione ambientale: da 700 hPa a 1060 hPa

Ciclo operativo

La lampada fotopolimerizzatrice è progettata per essere utilizzata per brevi periodi. Alla massima temperatura ambiente (32°C), 1 minuto di funzionamento per cicli consecutivi, 30 minuti di riposo (periodo di raffreddamento).

Risoluzione dei problemi
Se le soluzioni sotto riportate non risolvono il problema, contattare Ultradent al numero 800.552.5512. Al di fuori degli Stati Uniti, rivolgersi al distributore Ultradent oal proprio rivenditore di prodotti odontoiatrici.
ProblemaLa lampada non si accende

Possibili soluzioni
1) Premere il pulsante Tempo/Programma o il pulsante di accensione per riattivare la lampada dalla modalità di risparmio
energetico.
2) Controllare la spia rossa che indica batteria in esaurimento per verificare lo stato di carica della batteria.
3) Controllare che le batterie cariche siano correttamente inserite nell'unità.
4) Se lampeggiano i LED di avviso rossi e gialli significa che la temperatura interna della lampada ha raggiunto il limite
di sicurezza. Lasciare raffreddare la lampada fotopolimerizzatrice per 10 minuti o utilizzare un panno fresco umido perraffreddare rapidamente l'unità.
5) Se il LED rosso di avviso lampeggia e emette un segnale acustico continuo, chiamare il Servizio Clienti Ultradent per lariparazione.

Linee guida e dichiarazione del fabbricante per le emissioni elettromagnetiche
La lampada fotopolimerizzatrice è destinata all'uso nell'ambiente elettromagnetico specificato di seguito. Il cliente o l'utente deve garantire che sia utilizzata in unambiente di questo tipo.
Test di emissioni

Conformità

Ambiente elettromagnetico - Linee guida

Emissioni RF
CISPR 11

Gruppo 1

La lampada fotopolimerizzante è alimentata a batterie e non è influenzata da interferenze EMI,
RF o sovratensioni.

Emissione RF
CISPR 11Classe B

Emissioni armoniche
IEC 61000-3-2

N/A

La lampada fotopolimerizzante utilizza l’energia elettrica ed elettromagnetica solo per le funzioni
interne. Di conseguenza le emissioni RF sono molto basse e non sono tali da causare interferenze
con l’apparecchiatura elettrica che si trova nelle immediate vicinanze.

Fluttuazioni di tensione/
emissioni di flicker IEC
61000-3-3

N/A

La lampada fotopolimerizzante è adatta per l'uso in tutti i tipi di ambienti, compresi gli ambienti
domestici e quelli collegati direttamente alla rete di alimentazione pubblica a bassa tensione che
serve edifici adibiti ad uso residenziale.

Linee guida e dichiarazione del fabbricante per l’immunità elettromagnetica
La lampada fotopolimerizzatrice è destinata all'uso nell'ambiente elettromagnetico specificato di seguito. Il cliente o l'utente deve garantire che sia utilizzata in unambiente di questo tipo.

La lampada non rimane accesa per iltempo desiderato1) Controllare l’impostazione delle spie tempo e programma.
2) Controllare la spia della batteria in esaurimento per verificare lo stato di carica della batteria.
3) Controllare che le batterie cariche siano correttamente inserite nell'unità.

Test d’IMMUNITÀ

Livello di test IEC 60601

Livello di conformità

Linee guida per l’ambiente elettromagnetico

Scarica elettrostatica (ESD)
IEC 61000-4-2± 8 kV di contatto± 15 kV in aria

La lampada non polimerizza
correttamente le resine

1) Verificare che sulla lente non vi siano residui di resina o composito polimerizzati.
2) Indossando occhiali di protezione anti-UV di colore ambra adeguati, verificare che le spie a LED funzionino.
3) Controllare il livello d’intensità con un radiometro. Se si utilizza un radiometro, Ultradent raccomanda di controllare lalampada fotopolimerizzatrice impostando il programma potenza standard.
NOTA: Il valore numerico dell’intensità registrata può non corrispondere all’effettivo valore a causa dell’inadeguatezza dei
radiometri comuni e l’utilizzo di LED particolari nella lampada fotopolimerizzatrice. I radiometri differiscono notevolmente tradi loro e sono realizzati per specifiche lenti e puntali ottici.
4) Controllare la data di scadenza della resina o composito fotopolimerizzabile.
5) Assicurarsi di seguire una tecnica corretta in base alle raccomandazioni del fabbricante.

± 8 kV di contatto± 15 kV in aria

L'ambiente fisico dovrebbe essere limitato a quantosegue:
1) Codice IP: IP20
2) Non immergere in liquidi.
3) Non utilizzare in prossimità di gas infiammabili.
L'unità è di tipo non APG e non AP.
4) Intervallo di umidità per la conservazione:
10% - 95%
5) Intervallo di temperatura per la conservazione:
10 °C - 40 °C

Burst/transitori elettrici veloci
IEC 61000-4-4

Le batterie non si ricaricano

1) Accertarsi che le batterie siano inserite correttamente nel caricabatterie e caricarle da 1 a 3 ore.
2) Se le spie rosse sul caricabatterie non diventano verdi, chiamare il Servizio Clienti Ultradent per ordinare batterie e/ocaricabatterie di ricambio.
3) Se sul caricatore non sono visibili le spie verdi e nemmeno quelle rosse, chiamare il Servizio Clienti Ultradent per
ordinare o sostituire il caricabatterie e/o l’adattatore CA.

± 2 kV per linee di alimentazione
elettrica± 1 kV per linee di ingresso/uscita± 2 kV per linee di alimentazione
elettrica
Nota 1: La lampada
fotopolimerizzante non haporte I/O

La lampada fotopolimerizzante è alimentata abatteria e non può essere collegata alla rete dialimentazione CA.

Il caricabatterie non ricarica le batterie

1) Accertarsi che il caricabatterie sia collegato alla presa e che l’adattatore CA sia collegato ad una presa elettrica
funzionante.
2) Se le luci verdi o rosse sul caricabatterie non sono visibili, chiamare il Servizio Clienti Ultradent per avere un
caricabatterie e/o un adattatore CA nuovi.

Sovratensione
IEC 61000-4-5± 1 kV da linea a linea± 2 kV da linea a terra± 1 kV da linea a linea± 2 kV da linea a terra<5% U(>95% di calo in U per 0,5 cicli)<5% U(>95% di calo in U per 0,5 cicli)

Non è possibile modificare il
programma o gli intervalli di tempo

1) Tenere premuti i pulsanti Tempo/Programma e di accensione fino a quando una serie di segnali acustici indica che lalampada fotopolimerizzatrice è sbloccata.

Cali di tensione, brevi interruzioni
e variazioni delle linee in ingresso
dell’alimentazio-ne

40% U(60% di calo in U per 5 cicli)

40% U(60% di calo in U per 5 cicli)

70% U(30% di calo in U per 25 cicli)

70% U(30% di calo in U per 25 cicli)<5% U(>95% di calo in U per 5 s)

<5% U(>95% di calo in U per 5 s)
Nota 2: si ripristina
automaticamente

30 A/m30 A/mIEC 61000-4-11

Frequenza di rete
(50/60 Hz) campo magnetico
IEC 61000-4-8
48

Le prove di emissione armonica e fluttuazioni di tensione non sono applicabili alla lampada
fotopolimerizzante perché è alimentata a batteria.

Poiché la lampada fotopolimerizzante è alimentata
a batteria, non è soggetta a transienti elettrici,
sovratensioni, cali di tensioni, cortocircuiti, interruzioni
o variazioni della rete di alimentazione CA.
Il caricabatterie accessorio può essere soggetto
a quanto menzionato sopra, ma è separato dallalampada e non è essenziale per il funzionamento dellalampada fotopolimerizzante VALO senza fili.
Se la tensione della batteria della lampada
fotopolimerizzante scende a 4 VCC, l'unità non
funziona. La lampada fotopolimerizzante VALO
Senza fili si spegne. Una volta inserite le nuove
batterie e ripristinati i livelli di corrente corretti, lalampada fotopolimerizzante si riavvia e torna allo
stato precedente alla perdita di corrente. La lampada
fotopolimerizzante eseguirà un ripristino automatico
in caso di perdita di corrente.
I campi magnetici a frequenza di rete devono
trovarsi ai livelli caratteristici di una tipica ubicazione
all’interno di un ambiente residenziale, di assistenza
domestica, commerciale, ospedaliero o militarestandard.
49Linee guida e dichiarazione del fabbricante per l’immunità elettromagnetica

Linee guida e dichiarazione del fabbricante per le distanze di separazione raccomandate tra le apparecchiature di comunicazione RF portatili e mobili e la lampada
fotopolimerizzatrice VALO Cordless.

Nota 1: La lampada fotopolimerizzatrice non è dotata di porte o di linee I/O accessibili.

La lampada fotopolimerizzatrice deve essere utilizzata in un ambiente elettromagnetico in cui le interferenze da RF irradiate sono controllate. L'utente della lampada
fotopolimerizzatrice può contribuire a prevenire le interferenze elettromagnetiche rispettando la distanza minima fra gli apparecchi di comunicazione RF mobili e
portatili (trasmettitori) e la lampada fotopolimerizzatrice come indicato di seguito, in accordo con la potenza massima in uscita del dispositivo per le comunicazioni.

Linee guida e dichiarazione del fabbricante per l’immunità elettromagnetica dei sistemi di supporto non critici per la vita
La lampada fotopolimerizzatrice è destinata all'uso nell'ambiente elettromagnetico specificato di seguito. Il cliente o l'utente deve garantire che sia utilizzata in unambiente di questo tipo.
Test d’IMMUNITÀ

Conduzione RF

IEC 61000-4-6

Livello di test IEC
60601

3 Vrm

Da 150 kHz a 80 MHz

Livello di conformità

3 Vrm

Linee guida per l’ambiente elettromagnetico
Le apparecchiature per le comunicazioni RF portatili e mobili non devono essere utilizzate
ad una distanza da qualsiasi parte della lampada fotopolimerizzante, compresi i cavi,
inferiore alla distanza di separazione raccomandata, calcolata mediante l’equazione
applicabile alla frequenza del trasmettitore.
Distanza di separazione raccomandata:

Da 150 kHz a 80 MHz
da 80 MHz a 800 MHz

RF irradiata

3 V/m

3 V/m

da 800 MHz a 2,5 GHzP è la potenza massima di uscita del trasmettitore in watt (W) in base al produttore deltrasmettitore e d è la distanza di separazione raccomandata in metri (m).
Le resistenze di campo provenienti dai trasmettitori RF fissi, conformemente a quantodeterminato da una perizia elettromagnetica sul sitoa, dovrebbero essere inferiori al livellodi conformità di ogni gamma di frequenzeb.

IEC 61000-4-3

Da 80 MHz a 2,5 GHz

Da 80 MHz a 2,5 GHz

Distanza di separazione in base alla frequenza del trasmettitore(metri)
Potenza nominale massima in uscita
del trasmettitore(P in Watt

150 kHz – 80 MHz

80 MHz – 800 MHz

800 MHz – 2,5 GHz

0,01

0,12 metri

0,035 metri

0,07 metri

0,1

0,37 metri

0,11 metri

0,22 metri

1

1,7 metri

0,35 metri

0,7 metri

10

3,7 metri

1,11 metri

2,22 metri

100

11,7 metri

3,5 metri

7,0 metri

La lampada fotopolimerizzatrice è stata testata secondo IEC 60601-1-2:2014 e ha superato il test con le intensità di campo irradiato di 10 V/m tra 80 MHz e 2,5 GHz. Il
valore di 3Vrm corrisponde a V1 e il valore 10 V/m corrisponde a E1 nelle formule sopra riportate.
Per i trasmettitori valutati ad una potenza massima in uscita non elencata in precedenza, la distanza di separazione raccomandata d in metri (m) può essere stimata
utilizzando l'equazione applicabile alla frequenza del trasmettitore, dove P è la potenza massima in uscita del trasmettitore in watt (W) in base al produttore deltrasmettitore.
NOTA 1 A 80 MHz e 800 MHz, si applica la distanza di separazione per la gamma di frequenze più elevata.
NOTA 2 Le presenti linee guida potrebbero non essere applicabili in tutte le situazioni. La propagazione elettromagnetica è influenzata dall'assorbimento e dallariflessione di strutture, oggetti e persone.

Possono verificarsi interferenze nelle vicinanze dell'apparecchiatura contrassegnata dalseguente simbolo:

NOTA 1 A 80 MHz e 800 MHz, si applica la gamma di frequenze più elevata.
NOTA 2 Le presenti linee guida potrebbero non essere applicabili in tutte le situazioni. La propagazione elettromagnetica è influenzata dall'assorbimento e dallariflessione di strutture, oggetti e persone.
aNon è possibile prevedere teoricamente con precisione le intensità di campo per trasmettitori fissi, come le stazioni base per radiotelefoni (cellulari e senza fili) eradiomobili terrestri, apparecchi di radioamatori, trasmettitori radio in AM e FM e trasmettitori TV. Per valutare l'ambiente elettromagnetico creato da trasmettitori RF
fissi, si può considerare una perizia elettromagnetica del sito. Se l’intensità di campo misurata nel luogo in cui si usa la lampada fotopolimerizzante, supera il livellodi conformità RF applicabile di cui sopra, si dovrebbe porre sotto osservazione il funzionamento normale della lampada fotopolimerizzante VALO senza fili. Se si
osservano prestazioni anomale, possono essere necessarie misure aggiuntive, come il riorientamento o il riposizionamento della lampada fotopolimerizzatrice.
b
Oltre l'intervallo di frequenza compreso tra 150 kHz e 80 MHz, le intensità di campo dovrebbero essere inferiori a 3 V/m.

50

51

VALO™ GRAND

ESLÁMPARA DE FOTOCURADO
Descripción del producto

Con su espectro de banda ancha, la lámpara de polimerización VALO Grand está diseñada para polimerizar todos los productos fotopolimerizables en la gama de longitudes de onda de385-515 nm según la norma ISO 10650. La lámpara de curado VALO Grand utiliza las baterías recargables y el cargador de baterías VALO de Ultradent. La lámpara de fotopolimerización
está diseñada para descansar en un soporte de unidad dental estándar o puede montarse a medida utilizando el soporte incluido en el kit.

Componentes del producto VALO Grande:
•
•
•
•
•
•

Foto

Botones y luces

Nota:• Para evitar la contaminación cruzada y evitar que el material de composite dental se adhiera a la superficie de la lente y al cuerpo de la varilla, debe utilizarse una funda de barrera
aprobada por Ultradent sobre la lámpara de polimerización VALO en cada uso. Las fundas de barrera están pensadas para uso en un solo paciente.

Protector de luz para VALO Inalámbrica:

El Protector de luz para VALO Inalámbrica tiene forma ovalada, se puede girar para un uso máximo y se puede usar con una funda de barrera transparente.

LeyendaBotones de Encendido(Parte superior o inferior)
Luces de tiempo
Indicador de batería baja
Luz de Modo/Estado
Botón de Cambio de Modo/
Tiempo(Parte inferior)(Parte superior)Para todos los productos descritos, lea atentamente y comprenda todas las instrucciones y la información de la SDS antes de usar el producto.

Indicaciones de uso/Fin previsto

La fuente de iluminación para el fotocurado de materiales restauradores dentales y adhesivos fotoactivados.

Advertencias y precauciones
Grupo de riesgo 2
PRECAUCIÓN UV emitida por este producto. La exposición puede causar irritación de los ojos o la piel. Utilice una protección adecuada.
PRECAUCIÓN Radiación óptica posiblemente peligrosa emitida por este producto. No mire fijamente la lámpara durante su funcionamiento. Puede ser perjudicial para los ojos.
• NO mire directamente a la salida de luz. El paciente, el clínico y los asistentes deben llevar siempre protección ocular UV de color ámbar cuando se utilice la lámpara de curado.
• Para evitar el riesgo de descarga eléctrica, no se permite modificar este equipo. Utilice únicamente la fuente de alimentación y los adaptadores de enchufe Ultradent VALO incluidos.
Si estos componentes están dañados, no los utilice y llame al Servicio de Atención al Cliente de Ultradent para pedir un recambio.
• Los equipos portátiles de comunicaciones por RF pueden degradar el rendimiento si se utilizan a menos de 30 cm (12 pulg.).
• Utilice únicamente accesorios, cables y fuentes de alimentación autorizados para evitar un funcionamiento incorrecto, el aumento de las emisiones electromagnéticas o ladisminución de la protección electromagnética (consulte la sección Emisiones electromagnéticas).
• Para evitar el riesgo de incendio eléctrico asociado a la manipulación de las baterías:
⸰ NO mezcle baterías recargables con baterías no recargables u otros tipos de baterías.
⸰ NO intente cargar baterías no recargables. Cargar baterías no recargables puede provocar lesiones o daños materiales.
⸰ oNO esterilice en autoclave ni rocíe las baterías, los contactos de las baterías, el cargador o el adaptador de corriente alterna con líquido de ningún tipo. Si aparece
corrosión en los contactos del cargador de baterías, llame al servicio de atención al cliente de Ultradent para solicitar un recambio.
⸰ NO cargue las baterías cerca de materiales inflamables.
⸰ NO guarde el cargador en el quirófano clínico.
• Para evitar el riesgo de lesiones, NO utilice baterías que estén corroídas (oxidadas), abolladas, que desprendan olor o fluidos, que tengan el envoltorio roto o que les falte elenvoltorio, o que estén dañadas de cualquier otra forma. Llame al servicio de atención al cliente de Ultradent para solicitar baterías de repuesto.
• Utilice sólo las baterías recomendadas. Las baterías alternativas pueden provocar un mal funcionamiento.
• Para prevenir el riesgo de irritación o lesiones térmicas, evite los ciclos de polimerizado consecutivos y no exponga los tejidos blandos orales muy próximos durante más de 10
segundos en cualquier modo. Si se requieren tiempos de polimerizado más largos, utilice ciclos de polimerizado múltiples con periodos de descanso entre ciclos, o utilice unproducto de polimerización dual para evitar el calentamiento de los tejidos blandos.
• Tenga precaución al tratar a pacientes que sufran reacciones fotobiológicas adversas o sensibilizaciones, pacientes que estén recibiendo tratamiento de quimioterapia o pacientes en
tratamiento con medicación fotosensibilizante.
• Esta unidad puede ser susceptible a fuertes campos magnéticos o eléctricos estáticos, que podrían alterar la programación. Si sospecha que esto ha ocurrido, desenchufe la unidad
momentáneamente y vuelva a enchufarla a la toma de corriente.NO limpie la lámpara de curado con limpiadores cáusticos o abrasivos, ni la esterilice en autoclave, ni la sumerja
en ningún tipo de baño ultrasónico, desinfectante, solución limpiadora o líquido. Si no se siguen las instrucciones de procesamiento incluidas, la lámpara de polimerización puede
52

Instrucciones paso a paso
Preparación

1) Cargue las baterías antes de utilizar la lámpara de curado (consulte la sección Mantenimiento de las baterías).
2) Antes de cada uso, coloque una nueva funda de barrera sobre la lámpara de polimerización, minimizando las arrugas sobre la lente para obtener los mejores resultados.

1 - Lámpara de polimerización VALO Grand4 - Baterías recargables Ultradent VALO
1 - Cargador de batería Ultradent VALO con adaptador de corriente alterna de 12 VCC de calidad médica
1 - Paquete de muestras de fundas de barrera VALO
1 - Filtro de luz VALO Grand
1 - Soporte de montaje en superficie de la lámpara de polimerización con cinta adhesiva de doble cara

Resumen de los controles:

quedar inutilizada.
• Para evitar dañar el equipo, NO introduzca los dedos, instrumentos u otros objetos en el compartimento de las baterías de la lámpara de curado.
• Para evitar dañar el equipo, NO intente limpiar los contactos dorados, ni ninguna parte del compartimento de la pila. Llame al Servicio de Atención al Cliente de Ultradent si tiene
algún problema.
• Para evitar el riesgo de contaminación cruzada, las fundas de barrera son de un solo uso.
• Para reducir el riesgo de corrosión, retire la funda de barrera después de su uso.
• Para reducir el riesgo de que las resinas no se polimericen lo suficiente, no utilice la lámpara de polimerización si la lente está dañada.

Uso1) Cada modo de potencia se utiliza para el polimerizado de materiales dentales con fotoiniciadores. Consulte la Guía de Modo Rápido para conocer los tiempos de curado
recomendados.
NOTA: La lámpara de polimerización está programada para pasar secuencialmente de la Potencia Estándar a la Alta Potencia Plus y a la Potencia Xtra. Por ejemplo, para cambiar delmodo de potencia Estándar al modo de potencia Extra, es necesario pasar al modo de Alta Potencia Plus y después al modo de potencia Extra.
2) La lámpara de curado almacena el intervalo de tiempo y el modo utilizados más recientemente, y volverá a ellos por defecto cada vez que se cambien los modos o si se extraen laspilas.

Funcionamiento
MODO DE FOTOCURADO: Modo de potencia estándar

INTERVALOS DE TIEMPO: 5, 10, 15, y 20 segundos.
• La lámpara de fotocurado se establece de manera predeterminada en este modo cuando se enciende INICIALMENTE. La Luz de Modo/Estado estará en verde y las cuatro Luces deTiempo verdes se iluminarán, indicando el modo de Potencia Estándar.
• Para cambiar entre los intervalos de tiempo, presione rápidamente el botón de Tiempo/Modo.
• Presione el Botón de Encendido para iniciar el fotocurado. Para detener el fotocurado antes de completar un intervalo de tiempo, vuelva a presionar el botón de encendido.

MODO DE FOTOCURADO: Modo de Potencia Alta Plus

INTERVALOS DE TIEMPO: 1, 2, 3, y 4 segundos.
• Desde el modo de potencia estándar, mantenga presionado el botón de cambio de modo/tiempo durante 2 segundos y suéltelo. La luz de Modo/Estado será de color naranja, y las
cuatro luces de Tiempo verdes se iluminarán y parpadearán, indicando el modo de Alta Potencia.
• Para cambiar entre los intervalos de tiempo, presione rápidamente el botón de Tiempo/Modo.
• Presione el Botón de Encendido para fotocurar. Para detener el fotocurado, tras haber completado un intervalo de tiempo, presione nuevamente el Botón de Encendido.
• Para volver al modo de potencia estándar, mantenga presionado el botón de cambio de tiempo/modo durante 2 segundos y suéltelo, esto pasará al modo de potencia Xtra.
Mantenga presionado de nuevo durante 2 segundos, y suelte. La luz de Modo/Estado será verde y las cuatro luces de Tiempo verdes se iluminarán, indicando el modo Estándar.

MODO DE FOTOCURADO: Modo de Potencia Xtra

INTERVALO DE TIEMPO: Solo 3 segundos (Nota: El modo Potencia Xtra tiene un retraso de seguridad de 2 segundos al final de cada ciclo de fotocurado para limitar el calentamiento
durante el fotocurado consecutivo. Una vez terminado este retraso, un pitido le indicará que la unidad está lista para continuar con su uso).
• Desde el modo de potencia estándar, presione el botón de cambio de tiempo/modo durante 2 segundos, suelte, mantenga presionado nuevamente durante 2 segundos y suelte. La
luz de Modo/Estado será naranja y parpadeará, y tres de las luces de Tiempo verdes se iluminarán y parpadearán, indicando el modo de Potencia Xtra.
• Presione el Botón de Encendido para iniciar el fotocurado. Para detener el fotocurado, después de haber completado un intervalo de tiempo, presione nuevamente el botón deencendido.
• Para volver al modo de potencia estándar, mantenga presionado el botón de tiempo/modo durante 2 segundos y suéltelo. La luz de Modo/Estado será verde y las luces de Tiempo
verdes se iluminarán, indicando el modo de Potencia Estándar.

Modo de espera: La lámpara de fotocurado entrará en modo de espera tras 60 segundos de inactividad, y esto se indica mediante el parpadeo lento de la luz de modo/estado. Si
levanta o toca la unidad, se encenderá la luz de fotopolimerización y volverá automáticamente a la última configuración utilizada. Para maximizar la vida útil de la batería, no toque lalámpara de curado cuando no esté en uso.

Limpieza1) Deseche las fundas de barrera usadas en la basura normal después de cada paciente.
2) Véase la sección de procesamiento.

Instrucciones de montaje del soporte

1) El soporte debe montarse en una superficie plana y sin aceite.
2) Limpie la superficie con alcohol de quemar.
3) Despegue la cinta adhesiva del soporte.
4) Coloque el soporte de modo que la lámpara de curado se levante al retirarla. Presione firmemente en su sitio.

Guía rápida de modos:
Modo

Potencia Estándar
1000 mW/cm2

Potencia Alta Plus
1600 mW/cm2

Potencia Xtra
3200 mW/cm2
53

Mantenimiento de las pilas
Carga y cambio de pilas
La lámpara de curado se suministra con 4 pilas recargables de litio fosfato de hierro. Ultradent recomienda utilizar únicamente pilas recargables VALO.

Botón de encendido
Luces LED de Modo/Tiempo

Botones de tiempo

Opciones de tiempo

5s

10s

15s

20s

1s

2s

3s

4s

Solo 3s

Para cambiar el tiempo

Presione y suelte el botón de tiempo rápidamente para recorrer las opciones de tiempo.

Para cambiar el modo

Mantenga presionado el botón de tiempo durante 2 segundos y suéltelo. La lámpara de fotocurado pasará al siguiente modo.

LeyendaLEDs sólidos

LEDs parpadeantes

Guía rápida de fotocurado:
Modo de curado

Potencia* (mW)

Irradiancia* (mW/cm2)

Tiempo total de exposición (segundos)

Energía** (Julios)

Estándar

970

900

5/10/15/20

4.8J/9.7J/14.5J/19.4J

Alta potencia Plus

1,620

1,500

1/2/3/4

1,6J/3,2J/4,8J/6,5J

Extra

2,260

2,100

3

6.8J

*Salida nominal a una distancia de 2 mm desde la punta de la lente a la superficie superior del composite.
**Los valores de la sección de energía total por ciclo (julios) se han redondeado a la décima más próxima.

Guía rápida de advertencias:
Advertencia de nivel de potencia

Advertencia de temperatura

Advertencia de servicio

Advertencia LED

Cambiar pilas

Dejar enfriar

Llame a Servicio de atención al clientepara su reparación

Llame a Servicio de atención al clientepara su reparación• Batería baja: parpadeo lento• Apagado: 3 pitidos, parpadeo• No permite el funcionamiento• 3 pitidos• Parpadeo lento• No permite el funcionamiento• Sin sonido• Parpadeo cada 2 segundos• Permite el funcionamiento• 3 pitidos continuos• Un parpadeo rápido
• No permite el funcionamiento

Cómo cargar las pilas:1)
Coloque el cargador y las pilas en una zona donde no entren en contacto con líquidos o aerosoles desinfectantes.
2)
Introduzca las pilas en el cargador con el extremo positivo (+) apuntando hacia las luces indicadoras del cargador.
3)
Las luces verdes indican que las pilas están listas para su uso.
4)
Las pilas tardarán entre 1 y 3 horas en cargarse. Deje las pilas en el cargador hasta que estén listas para su uso.
NOTA: Si la luz roja del cargador no se vuelve verde cuando las pilas llevan más de tres horas cargándose, es posible que la pila esté degradada (dañada) y no pueda cargarse. Pruebe
con una pila nueva o llame al servicio de atención al cliente de Ultradent para pedir un nuevo juego de pilas recargables.
Cómo cambiar/insertar las pilas:
1)
Retire la tapa trasera girando un cuarto de vuelta en sentido antihorario.
Retire las pilas.
2)
3)
Inserte primero el lado positivo (+) de las pilas nuevas.
Vuelva a colocar la tapa trasera alineándola y empujando suavemente mientras gira en el sentido de las agujas del reloj. La tapa hará clic cuando esté completamente
4)acoplada.
5)
La unidad está lista para su uso.
Pilas bajas: la luz de curado indica al usuario que ha llegado el momento de cambiar las pilas cuando el indicador luminoso de pilas bajas parpadea en rojo. Si la carga de las pilas es
demasiado baja, se emitirá un sonido de advertencia de 3 pitidos y la lámpara de curado no permitirá seguir funcionando hasta que se recarguen las pilas o se introduzcan pilas nuevas.
(Consulte la guía de advertencia rápida)
Tiempo de carga y duración de la pila: la duración de la batería completamente cargada en la lámpara de curado depende del intervalo Modo/Tiempo, del tipo de pilas, de la cantidad
de uso y de la eficiencia del LED.
• Intervalo de recarga recomendado: recargue cuando sea necesario o, cuando se encienda el indicador de batería baja. La esperanza de vida depende del uso.
• Pilas adicionales: la lámpara de curado viene con 4 pilas recargables.
Reparación del fabricante1) Las reparaciones solo deben ser realizadas por personal de servicio autorizado. Ultradent debe proporcionar al personal de servicio la documentación necesaria para realizar las
reparaciones.
2) Cuando envíe las unidades para su reparación, mantenimiento o calibración, retire siempre las pilas de la lámpara de curado y del cargador. Envuelva las pilas, el cargador, el
adaptador y la lámpara de curado por separado en la caja de devolución.
3) Envíe las pilas de acuerdo con la normativa local.
Garantía
Ultradent Products, Inc. («Ultradent») garantiza que este producto, durante un periodo de 5 años a partir de la fecha de compra, cuando se utilice de acuerdo con las instrucciones de
uso incluidas con el producto, (i) se ajustará en todos los aspectos materiales a las especificaciones establecidas en la documentación de Ultradent que acompaña al producto; y (ii)estará libre de defectos materiales y de fabricación.
Esta garantía limitada es intransferible y se aplica únicamente al comprador original y no se extiende a los propietarios posteriores del producto. Esta garantía limitada no cubre ningúnotro componente accesorio como, por ejemplo, pilas, cargadores, adaptadores u objetivos adaptables. Esta garantía limitada quedará anulada si el producto falla o se daña debido
a negligencia, abuso, uso indebido, accidente, modificación, manipulación, alteración o incumplimiento de las instrucciones de uso aplicables. A modo de ejemplo únicamente, un
producto que se caiga y se dañe no estará cubierto por esta garantía. Para poder acogerse a esta garantía limitada, deberá enviarse a Ultradent una prueba de compra (por ejemplo, el
recibo de compra o documentación similar) junto con el producto defectuoso.
Un producto defectuoso que cumpla las condiciones de garantía aquí establecidas será, a discreción exclusiva de Ultradent, reparado o sustituido. En ningún caso la responsabilidad deUltradent por el producto excederá el precio de compra pagado por el comprador. Bajo ninguna circunstancia Ultradent será responsable de daños indirectos, incidentales, previstos,
imprevistos, especiales o consecuentes que surjan de o en relación con el uso de este producto.

Procesamiento

Este producto está destinado a ser utilizado con una funda de barrera. Véase más arriba el mantenimiento y la limpieza general.

Almacenamiento y eliminación

Si almacena la lámpara de polimerización durante períodos de más de 2 semanas o la empaca para viajar, retire siempre las pilas. Si las pilas se dejan en la unidad durante largosperíodos de tiempo sin recargarlas, pueden volverse no funcionales o imposibles de cargar. No almacene las pilas a temperaturas superiores a los 40° C (104° F) ni a la luz solar directa.

Mantenimiento

Limpieza general de la lámpara de curado
Después de cada uso, humedezca una gasa o un paño suave con un desinfectante de superficies aprobado y limpie la superficie y la lente. Los limpiadores no autorizados pueden dañar
la lámpara de curado.
LIMPIADORES ACEPTABLES:•		•		

Almacenamiento y transporte de la lámpara de fotocurado:• Temperatura: +10° C a + 40° C (+50° F a + 104° F)• Humedad relativa: 10% a 95%• Presión ambiental: 500 hPa a 1060 hPa
Cuando deseche residuos electrónicos (es decir, lámparas de fotocurado, cargadores, baterías/pilas y fuentes de alimentación), siga las normas locales sobre residuos y reciclaje.

Alcohol isopropílico al 70 %Etanol al 70 %Limpieza general del escudo de luz:
Limpie el protector de luz VALO Cordless utilizando cualquier desinfectante de superficies. NO esterilizar en autoclave.
Reparación
Mantenimiento realizado por el usuario
1)
Utilice una funda de barrera para evitar que el composite dental se adhiera a la superficie de la lente. Si es necesario, utilice un instrumento dental de plástico o aceroinoxidable para retirar con cuidado el composite adherido. No utilice herramientas que puedan dañar el objetivo.
2)
Los fotómetros difieren mucho y están diseñados para puntas de guía de luz y objetivos específicos. Ultradent recomienda comprobar rutinariamente la salida en elmodo de potencia estándar. NOTA: la salida numérica real estará sesgada debido a la inexactitud de los medidores de luz comunes y al paquete de LED personalizados de la lámpara de
curado.
54

Consideraciones técnicas
Accesorios
Elemento

Información CE

Protector de luz para VALO
Inalámbrica

55

Elemento

Información CE

Información técnica/datos

Bandas de longitud de onda de fotocurado de compuesto efectivas:

Cargador VALO

POTENCIA RADIANTE ESPECTRAL (mW/nm)

BANDAS DE LONGITUD DE ONDA EFECTIVAS PARA EL CURADO DE COMPOSITES(Rango de longitud de onda utilizable: 385-515 nm)

Pilas VALO

Efectivo inferior
banda de longitud
de onda395-415 nm

Banda de longitud
de onda superior
efectiva
440-480 nm

Extra
Potencia Alta
EstándarLONGITUD DE ONDA (nm)Atributo

Información/Especificación

Lente

Diámetro 11.7 mm

Rango de longitud
de onda• Rango de longitud de onda utilizable: 385 - 515nm• Longitud de onda máxima: 395 - 415nm y 440 - 480nm
Tabla de comparación de la radiación saliente nominal
Instrumento de
medidaTabla de intensidad
de la luz

56

‡ Analizador de espectro de gigahercios
SalidaPotencia total

Abertura demedidor

15 mm

15 mm

Potencia Estándar
(±10%)

900 mW/cm²

970 mW

Potencia Alta Plus(±10%)

1500 mW/cm²

1615 mW

Potencia Xtra(±10%)

2100 mW/cm²

La Exitancia Radiante variará en función dela capacidad del instrumento, el método de
medición y la colocación de la luz.
‡ La exitancia radiante cumple la norma ISO
10650 cuando se mide con un analizador deespectro Gigahertz-Optik.

2260 mW

Lámpara defotocurado VALO
Inalámbrica

Clasificaciones: IEC 60601-1 (Seguridad), IEC
60601-1-2 (EMC)

Peso:• Con pilas: 6 oz. (190 gramos)• Sin pilas: 5 oz. (150 gramos)
Dimensiones: (8 x 1.28 x 1.06) pulgadas, (203 x 32.5 x 27) mm

Fuente dealimentación delcargador

Salida - 12VDC a 500mA
Entrada: 100VAC a 240VAC
Ultradent P/N 5930 Fuente de alimentación delcargador VALO con enchufes internacionales

Clasificaciones: IEC 60601-1 (Seguridad)
Longitud del cable - 6 pies (1.8 metros)
La fuente de alimentación del cargador VALO es una fuente dealimentación de grado médico clase II y proporciona aislamiento de la redeléctrica PRINCIPAL

57

Información variada

Atributo

Información/Especificación

Cargador VALO

Cargador inteligente VALO de pilas de fosfato de litio y hierro de 3.6VCC:• Apagado automático cuando las pilas están completamente cargadas• Detección automática de pilas defectuosas• Protecciones: Térmica, sobrecarga, cortocircuito, polaridad inversa
⸰ LED rojo - Cargando
⸰ LED verde - Vacío o completamente cargado
⸰ LED apagado - Cortocircuito• Tiempo de carga: De 1 a 3 horas
Clasificación: CE, WEEE

Pilas VALO

Recargables: Composición química segura de fosfato de hierro y litio (LiFePO4) RCR123A• Tensión de funcionamiento: 3.2VCC
Clasificaciones: UL, CE, RoHS, WEEE

Condiciones defuncionamiento

Temperatura: +10° C a +32° C (+50° F a +90° F)
Humedad relativa: 10% a 95%
Presión ambiental: 700 hPa a 1060 hPa

Ciclo de trabajo

La lámpara de fotocurado está diseñada para un funcionamiento a corto plazo. A la temperatura ambiente máxima (32° C) 1
minuto de ciclo de espalda con espalda, 30 minutos de apagado (período de enfriamiento).

Solución de problemas

La lámpara de fotocurado está diseñada para su uso en el entorno electromagnético especificado a continuación. El cliente o usuario debe asegurarse de usarla en un
entorno de tales características.
Prueba de emisiones

Cumplimiento

Entorno electromagnético - guía

Emisiones de RF
CISPR 11

Grupo 1

La lámpara de fotocurado utiliza pilas y no se ve afectada por EMI, RF o supresión desobretensiones.

Emisión de RF
CISPR 11

Clase B

HEmisiones de armónicos
IEC 61000-3-2

N/A

La lámpara de fotocurado utiliza energía eléctrica y electromagnética solo para sus funciones
internas. Por lo tanto, las emisiones de RF son muy bajas y es probable que no causeninterferencias en los equipos electrónicos cercanos.

Fluctuaciones de tensión/
emisiones de parpadeo IEC
61000-3-3

N/A

La emisión de armónicos y las pruebas de fluctuación de tensión no son aplicables a la lámpara defotocurado porque está alimentada por pilas.
La lámpara de polimerización es adecuada para su uso en todos los establecimientos, incluidos
los establecimientos domésticos y los conectados directamente a la red pública de suministro deenergía de baja tensión que suministra a los edificios para uso doméstico.

Guía y declaración del fabricante para inmunidad electromagnética

Si las soluciones que se sugieren a continuación no rectifican el problema, por favor, póngase en contacto con Ultradent en el 800.552.5512. Fuera de los Estados
Unidos, llame a su distribuidor de Ultradent o a su distribuidor dental.
Problema

Guía y declaración del fabricante para emisiones electromagnéticas

Posibles soluciones

La lámpara no se enciende

1) Presione el botón de tiempo/modo o el botón de encendido para activarla desde el modo de ahorro de energía.
2) Compruebe el indicador rojo de batería baja para ver el estado de carga de la batería.
3) Verifique que las pilas nuevas estén insertadas correctamente en la unidad.
4) Si los LED de advertencia rojos y amarillos parpadean, significa que la lámpara de fotocurado ha alcanzado su límite deseguridad de temperatura interna. Deje que la lámpara de fotocurado se enfríe durante 10 minutos o use una toalla
húmeda y fría para enfriar la unidad rápidamente.
5) Si el LED de advertencia rojo parpadea y suena continuamente, llame al Servicio al cliente de Ultradent para solicitar
una reparación.

La luz no se mantiene encendida
durante el tiempo deseado

1) Compruebe el modo y las luces de sincronización para la entrada de tiempo correcta.
2) Verifique el indicador de batería baja para ver el estado de carga de la batería.
3) Verifique que las pilas nuevas estén correctamente insertadas en la unidad.

La lámpara no fotocura las resinas
correctamente.

1) Revise la lente en busca de resinas/compuestos fotocurados residuales.
2) Utilizando la protección ocular contra radiación UV color ámbar, verifique que las luces LED estén funcionando.
3) Verifique el nivel de potencia con un fotómetro. Si usa un medidor de luz, Ultradent recomienda verificar la lámpara defotocurado en el modo de Potencia Estándar.
NOTA: La salida numérica verdadera estará sesgada debido a la inexactitud de los medidores de luz comunes y el paquete de
LED personalizado que utiliza la lámpara de fotocurado. Los fotómetros difieren ampliamente entre sí y están diseñados para
puntas guía de luz y lentes específicas.
4) Verifique la fecha de caducidad de la resina fotocurable.
5) Asegúrese de seguir la técnica adecuada de acuerdo con las recomendaciones del fabricante.

Las pilas no se cargan

1) Asegúrese de que las pilas estén insertadas en el cargador en la orientación correcta y permita que las pilas se carguen
durante 1 a 3 horas.
2) Si las luces rojas del cargador no cambian a verde, llame al Servicio de atención al cliente de Ultradent para pedir pilasde repuesto y/o un cargador nuevo.
3) Si no se ven luces verdes ni rojas en el cargador, llame al Servicio al cliente de Ultradent para pedir o reemplazar elcargador y/o el adaptador de CA.

El cargador no carga las pilas

1) Asegúrese de que el cargador esté enchufado y que el adaptador de CA esté enchufado a una toma de corriente que
funcione.
2) Si las luces verdes o rojas del cargador no están visibles, llame al Servicio al cliente de Ultradent para obtener un nuevo
cargador y/o adaptador de CA.

No se puede cambiar el modo o losintervalos de tiempo

1) Mantenga presionados los botones Tiempo/Modo y Encendido hasta que una serie de pitidos indique que la lámpara defotocurado está desbloqueada.

La lámpara de fotocurado está diseñada para su uso en el entorno electromagnético especificado a continuación. El cliente o usuario debe asegurarse de usarla en
un entorno de tales características.
Prueba de INMUNIDAD

Nivel de prueba IEC 60601

Nivel de cumplimiento

Guía sobre el Entorno Electromagnético

Descarga electrostática (ESD)± 8 kV contacto± 15 kV aire

± 8 kV contacto± 15 kV aire

El entorno físico debe limitarse a lo siguiente:
1) Código IP: IP20
2) No sumergir en líquidos.
3) No usar cerca de gas inflamable. La unidad es
no APG y no AP.
4) Límites de humedad del almacenamiento:
10% - 95%
5) Límites de temperatura de almacenamiento:
10° C - 40° C

Ráfaga/transitorios eléctricos rápidos
IEC 61000-4-4± 2 kV para las líneas desuministro de alimentación± 1 kV para líneas de entrada/
salida± 2 kV para las líneas desuministro de alimentación
Nota 1: La lámpara de fotocuradono tiene puertos E/S

La lámpara de fotocurado funciona con pilas y no
puede conectarse a la red eléctrica principal CA.

Sobrecarga
IEC 61000-4-5± 1 kV línea a línea± 2 kV línea a tierra± 1 kV línea a línea± 2 kV línea a tierra

IEC 61000-4-2

Tensión, caídas, cortocircuitos,<5% U
interrupciones y variaciones en las (>95% de caída en U paralíneas de entrada del suministro
0.5 ciclo)
de alimentación
40% U
IEC 61000-4-11(60% de caída en U para 5 ciclos)<5% U(>95% de caída en U para
0.5 ciclo)

Frecuencia de energía(50/60 Hz) campo magnético

30 A/m

40% U(60% de caída en U para 5 ciclos)

Debido a que la lámpara de fotocurado funciona con
pilas, no está sujeta a transitorios eléctricos, aumentos
de tensión, a caídas de tensión, a cortocircuitos, ainterrupciones ni a variaciones en la alimentación de
CA PRINCIPAL.
El cargador de batería accesorio puede estar sujetoa lo anterior, pero está separado y no es crítico parael funcionamiento de la lámpara de fotocurado VALO
Inalámbrica.

Si la tensión de la pila de la lámpara de fotocurado
cae a 4VCD, la unidad no podrá funcionar. La lámpara
70% U
70% U
de fotocurado VALO Inalámbrica se apagará. Cuando(30% de caída en U para 25 ciclos) (30% de caída en U para 25 ciclos) se colocan pilas nuevas y se restauran los niveles
de energía adecuados, la lámpara de fotocurado
<5% U
<5% U
se reiniciará y volverá al mismo estado anterior a la(>95% de caída en U para 5 s)(>95% de caída en U para 5 s)
pérdida de energía. La lámpara de fotocurado se
Nota 2: Se recupera
recuperará automáticamente en caso de pérdida
automáticamente
de energía.
30 A/mIEC 61000-4-8

Los campos magnéticos de la frecuencia energética
deben estar en niveles característicos de una
ubicación típica en un entorno residencial, deatención médica a domicilio, comercial, hospitalarioo militar típico.

Nota 1: Lámpara de fotocurado no está equipada con ningún puerto ni con ninguna línea de E/S accesible.
58

59

Guía y declaración del fabricante para inmunidad electromagnética para sistemas de soporte no vitales
La lámpara de fotocurado está diseñada para su uso en el entorno electromagnético especificado a continuación. El cliente o usuario debe asegurarse de usarla en
un entorno de tales características.

Guía y declaración de fabricación para las distancias de separación recomendadas entre equipos de comunicaciones de RF portátiles y móviles y la lámpara defotocurado VALO Inalámbrica.

Prueba de
INMUNIDAD

La lámpara de fotocurado está diseñada para su uso en un entorno electromagnético en el que se controlan las perturbaciones de RF irradiadas. El usuario dela lámpara de fotocurado puede ayudar a prevenir la interferencia electromagnética manteniendo una distancia mínima entre los equipos de comunicaciones
de RF portátiles y móviles (transmisores) y la lámpara de fotocurado según se recomienda a continuación, según la potencia de salida máxima del equipo decomunicaciones.

Conducción de RF

IEC 61000-4-6

Nivel de prueba IEC
60601

3 Vrms150 kHz a 80 MHz

Nivel de cumplimiento Guía sobre el Entorno Electromagnético

3 Vrms150 kHz a 80 MHz

El equipo de comunicaciones de RF portátil y móvil no debe utilizarse más cerca de ninguna
parte de la lámpara de fotocurado, incluidos los cables, que la distancia de separación
recomendada calculada a partir de la ecuación aplicable a la frecuencia del transmisor.
Distancia de separación recomendada:

3 V/m

3 V/m

P es la clasificación de potencia máxima de salida del transmisor en vatios (W) según elfabricante del transmisor y d es la distancia de separación recomendada en metros (m).
Las intensidades de campo de los transmisores de RF fijos, según lo determinado por
un estudioa de emplazamientos electromagnéticos, deben ser menores que el nivel decumplimiento en cada intervalob de frecuencia.

IEC 61000-4-3

80 MHz a 2.5 GHz

80 MHz a 2.5 GHz

Se pueden producir interferencias cerca del equipo marcado con el siguiente símbolo:

NOTA 1: a los 80 MHz y 800 MHz, se aplica el intervalo de frecuencia más alto.
NOTA 2: estas pautas podrían no aplicarse en todas las situaciones. La propagación electromagnética se ve afectada por la absorción y la reflexión de las estructuras,
objetos y personas.
a
Las intensidades de campo de los transmisores fijos, como las estaciones base para los teléfonos de radio (celulares/inalámbricos) y las radios móviles terrestres,
las emisoras de radioaficionados, las transmisiones de radio AM y FM y las transmisiones de televisión no se pueden predecir teóricamente con precisión. Paraevaluar el entorno electromagnético debido a los transmisores de RF fijos, debe considerarse un estudio electromagnético del lugar. Si la intensidad de campo
medida en la ubicación en la que se usa la lámpara de fotocurado excede el nivel de cumplimiento de RF aplicable anterior, debe observarse la lámpara defotocurado VALO Inalámbrica para verificar el funcionamiento normal. Si se observa un funcionamiento anormal, pueden ser necesarias medidas adicionales, como
reorientar o reubicar la lámpara de fotocurado.
b
En el intervalo de frecuencia de 150 kHz a 80 MHz, las intensidades de campo deben ser inferiores a 3 V/m.

60

Potencia nominal de salida máxima
del transmisor(P en vatios)

150 kHz – 80 MHz

80 MHz – 800 MHz

800 MHz – 2.5 GHz

0.01

0.12 metros

0.035 metros

0.07 metros

0.1

0.37 metros

0.11 metros

0.22 metros

1

1.7 metros

0.35 metros

0.7 metros

10

3.7 metros

1.11 metros

2.22 metros

100

11.7 metros

3.5 metros

7.0 metros

80 MHz a 800 MHz

800 MHz a 2.5 GHz
RF irradiadas

Distancia de separación según la frecuencia del transmisor(metros)

La lámpara de fotocurado ha sido probada de acuerdo con la norma IEC 60601-1-2: 2014 y se sometió a intensidad de campo radiada de 10 V/m entre 80 MHz y 2.5
GHz. El valor de 3Vrms corresponde a V1 y el valor 10 V/m corresponde a E1 en las fórmulas anteriores.
Para los transmisores clasificados con una potencia de salida máxima no enumerada anteriormente, la distancia de separación recomendada d en metros (m) se
puede calcular utilizando la ecuación aplicable a la frecuencia del transmisor, donde P es la clasificación de potencia máxima de salida del transmisor en vatios (W)según el fabricante del transmisor.
NOTA 1: a 80 MHz y 800 MHz, se aplica la distancia de separación para el intervalo de frecuencia más alto.
NOTA 2: estas pautas podrían no aplicarse en todas las situaciones. La propagación electromagnética se ve afectada por la absorción y la reflexión de las estructuras,
objetos y personas.

61

VALO™ GRANDPT

FOTOPOLIMERIZADOR
Descrição do Produto

Com o seu espectro de banda larga, a luz de polimerização VALO Grand está concebida para polimerizar todos os produtos fotopolimerizáveis no intervalo de comprimento de onda
de 385-515 nm de acordo com a ISO 10650. A luz de polimerização VALO Grand utiliza as baterias recarregáveis e o carregador de bateria VALO da Ultradent. O fotopolimerizador foi
projetado para ser colocado em um suporte de unidade odontológica padrão ou pode ser montado de forma personalizada usando o suporte incluído no kit.

Componentes de Produto VALO Grande:
•
•
•
•
•
•

FotografiaBotões e indicadores luminosos

Nota:• Para ajudar a evitar a contaminação cruzada e evitar que o compósito dentário adira à superfície da lente ou do corpo da luz de polimerização, deve ser utilizada uma mangabarreira aprovada pela Ultradent sobre a luz de polimerização VALO em cada utilização. As mangas barreira destinam-se à utilização num único paciente.

Protetor de luz VALO Sem fio:

O protetor de luz VALO Sem fio é oval, pode ser girado para uso máximo e pode ser usado com uma luva de barreira transparente.

LegendaBotão(ões) Potência
(Topo ou fundo)
Luzes de temporização
Bateria baixa – Indicador
Luz de Modo/EstadoBotão de temporização/Mudança de modo

(Base)(Topo)

Para todos os produtos descritos, leia atentamente e compreenda todas as instruções e informações nas FDS antes de usar.

Indicações de Uso/Finalidade Pretendida

A fonte de iluminação para a polimerização de materiais de restauração dentária fotoativados e adesivos.

Avisos e Precauções
Grupo de Risco 2
CUIDADO UV emitido por este produto. A exposição pode provocar Irritação nos olhos ou na pele. Use uma proteção apropriada.
CUIDADO Este produto emite radiações ópticas possivelmente perigosas. Não olhe para a lâmpada em funcionamento. Pode ser prejudicial para os olhos.
• NÃO olhe diretamente para a saída de luz. O doente, o médico e os assistentes devem sempre usar proteção ocular de cor laranja quando a luz de polimerização estiver a serutilizada.
• Para evitar o risco de choque elétrico, não é permitida qualquer modificação deste equipamento. Utilize apenas a fonte de alimentação Ultradent VALO e os adaptadores de tomada
incluídos. Se estes componentes estiverem danificados, não os utilize e ligue para o Serviço de Atendimento ao Cliente Ultradent para solicitar uma substituição.
• Os equipamentos portáteis de comunicação RF podem degradar o desempenho se forem utilizados a uma distância superior a 30 cm (12 pol.).
• Utilize apenas acessórios, cabos e fontes de alimentação autorizados para evitar operações inadequadas, aumento das emissões eletromagnéticas ou diminuição da imunidade
eletromagnética (consulte a secção Emissões Eletromagnéticas).
• Para evitar o risco de incêndio elétrico associado ao manuseamento de baterias:⸰ NÃO misture pilhas recarregáveis com pilhas não recarregáveis ou outros tipos de pilhas.
⸰ NÃO tente carregar pilhas não recarregáveis. Carregar pilhas não recarregáveis pode resultar em lesões ou danos materiais.
⸰ NÃO faça autoclave ou borrife pilhas, contactos de pilha, carregador ou adaptador de energia CA com qualquer tipo de líquidos. Se surgirem sinais de corrosão nos contactos docarregador de bateria, ligue para o Serviço de Atendimento ao Cliente Ultradent para solicitar uma substituição.
⸰ NÃO carregue pilhas em redor de materiais inflamáveis.
⸰ NÃO mantenha o carregador em operação clínica.
• Para evitar o risco de ferimentos, NÃO utilize pilhas que estejam corroídas (ferrugem), amolgadas, emitam odores ou líquidos, tenham um invólucro rasgado ou ausente, ou que
estejam danificadas de qualquer outra forma. Ligue para o Serviço de Atendimento ao Cliente da Ultradent para solicitar baterias de substituição.
• Utilize apenas as pilhas recomendadas. Pilhas alternativas podem causar mau funcionamento.
• Para evitar o risco de irritação térmica ou lesão, evite ciclos de polimerização consecutivos e não exponha os tecidos moles orais na proximidade por mais de 10 segundos em
qualquer modo. Se forem necessários tempos de polimerização mais longos, alterne ciclos de polimerização com períodos de descanso ou utilize um produto de polimerização dupla
para evitar o aquecimento dos tecidos moles.
• Tome precaução ao tratar doentes que sofrem de reações fotobiológicas adversas ou sensibilidades, doentes que estão em tratamento de quimioterapia ou doentes em tratamento
com medicação fotossensibilizante• Esta unidade pode ser suscetível a fortes campos magnéticos ou elétricos estáticos, o que poderá perturbar a programação. Se suspeitar que isso ocorreu, desligue a unidade
momentaneamente e em seguida ligue-a novamente à tomada.
• NÃO limpe a luz de polimerização com produtos cáusticos ou abrasivos, autoclave ou mergulhe em qualquer tipo de banho de ultrassons, desinfetantes, soluções de limpeza ou
62

Instruções passo a passo
Preparação

1) Carregue as pilhas antes de utilizar a luz de polimerização (Consulte a secção Manutenção da bateria).
2) Antes de cada utilização, coloque uma nova manga barreira sobre a luz de polimerização, minimizando as rugas sobre a lente para obter melhores resultados.

1 – Luz de polimerização VALO Grand4 – Baterias recarregáveis Ultradent VALO
1 – Carregador de bateria Ultradent VALO com adaptador de energia CA de 12 VCC de nível médico
1- Pacote de amostras de manga de barreira VALO
1- Escudo protetor de luz sem fio VALO.
1 – Suporte de montagem em superfície do fotopolimerizador com fita adesiva dupla face

Vista geral dos comandos:líquidos. O incumprimento das instruções de processamento incluídas pode tornar a luz de polimerização inoperável.
• Para evitar danificar o equipamento, NÃO insira dedos, instrumentos ou outros objetos no compartimento da pilha da luz de polimerização
• Para evitar danificar o equipamento, NÃO tente limpar os contactos dourados, ou qualquer parte do compartimento da pilha. Ligue para o Serviço de Atendimento ao Cliente da
Ultradent se houver alguma preocupação.
• Para evitar o risco de contaminação cruzada, as mangas barreira são de utilização única para o doente.
• Para reduzir o risco de corrosão, remova a manga de barreira após a utilização.
• Para reduzir o risco de resinas mal polimerizadas, não utilize a luz de polimerização se a lente estiver danificada

Uso1) Cada modo de energia é usado para a polimerização de materiais dentários com fotoiniciadores. Consulte o Guia do Modo Rápido para saber os tempos de polimerização
recomendados.
NOTA: O fotopolimerizador está programado para mudar do modo Potência Standard para o modo Alta Potência Plus e para o modo Potência Xtra em sequência. Por exemplo, paramudar do modo Potência Standard para o modo Potência Xtra é necessário passar para o modo Alta Potência Plus e depois para o modo Potência Xtra.
2) O fotopolimerizador guarda sempre o último intervalo de tempo utilizado e o modo, e volta para a predefinição desse intervalo de tempo sempre que os modos forem alterados ou
se as baterias forem removidas.

Operação
MODO DE POLIMERIZAÇÃO: Modo Potência Standard

INTERVALOS DE TEMPO: 5, 10, 15, e 20 segundos.
• O fotopolimerizador volta para este modo no arranque INICIAL. A luz de Modo/Estado ficará verde e as quatro luzes verdes de temporização se acenderão, indicando o modoPotência Standard.
• Para alterar os intervalos de tempo, pressione rapidamente o botão Mudar Tempo/Modo• Pressione o botão de Potência para polimerizar. Para parar a polimerização antes da conclusão de um intervalo de tempo, pressione novamente o botão de Potência.

MODO DE POLIMERIZAÇÃO: Modo Alta Potência Plus

INTERVALOS DE TEMPO: 1, 2, 3, e 4 segundos.
• No modo Potência Standard, pressione o botão Mudar Tempo/Modo durante 2 segundos soltando em seguida. A luz de Modo/Estado ficará laranja e as quatro luzes verdes detemporização se acenderão piscando, indicando o modo Alta Potência.
• Para alterar os intervalos de tempo, pressione rapidamente o botão Mudar Tempo/Modo• Pressione um dos botões de Potência para polimerizar. Para parar a polimerização antes da conclusão de um intervalo de tempo, pressione novamente o botão de Potência.
• Para voltar ao modo Potência Standard, pressione o botão Mudar Tempo/Modo durante 2 segundos e solte-o, isto mudará para o modo Potência Xtra. Pressione novamente por 2
segundos e solte. A luz de Modo/Estado ficará verde e as quatro luzes verdes de temporização se acenderão, indicando o modo Potência Standard.

MODO DE POLIMERIZAÇÃO: Modo Potência Xtra

INTERVALO DE TEMPO: apenas 3 segundos (Nota: O modo Potência Xtra tem um atraso de segurança de 2 segundos no fim de cada ciclo para limitar o aquecimento durante a
polimerização consecutiva. No final do atraso soará um bip indicando que a unidade está pronta para utilização).
• No modo Potência Standard, pressione o botão Mudar Tempo/Modo durante 2 segundos, solte-o, pressione novamente durante 2 segundos e solte-o. A luz de Modo/Estado ficarálaranja piscando e as três luzes verdes de temporização se acenderão piscando, indicando o modo Potência Xtra.
• Pressione o botão de Potência para polimerizar. Para parar a polimerização antes da conclusão de um intervalo de tempo, pressione novamente o botão de Potência.
• Para voltar ao modo Potência Standard, pressione o botão mudar Tempo/Modo durante 2 segundos e solte-o. A luz de Modo/Estado ficará verde e as luzes verdes de temporização
se acenderão, indicando o modo Potência Standard.

Modo Suspensão: O fotopolimerizador passará para o Modo de Suspensão após 60 segundos de inatividade, conforme indicado por um piscar lento da luz de modo/estado. Levantar
ou tocar na unidade despertará o fotopolimerizador e fará com que volte automaticamente à última configuração usada. Para maximizar a duração da bateria, não perturbe o
fotopolimerizador quando não estiver em uso.

Limpeza

1) Descarte as mangas de barreira usadas com o lixo normal após seu uso com cada paciente.
2) Veja Seção de Processamento.

Instruções para o suporte de montagem

1) O suporte deve ser montado numa superfície plana e livre de óleo.
2) Limpe a superfície com álcool.
3) Retire a película da fita adesiva do suporte.
4) Posicione o suporte de modo que o fotopolimerizador se desloque para cima quando retirado. Pressione firmemente até ficar no lugar.

63

Guia Rápido de Modos:Potência Standard
1000 mW/cm2

Modo

Alta Potência Plus
1600 mW/cm2

Potência Xtra
3200 mW/cm2

Botão Potência
LEDs Modo/Tempo

1) Use uma luva de barreira para evitar que o compósito dentário adira à superfície da lente. Se necessário, utilize um instrumento dentário de plástico ou aço inoxidável para removercuidadosamente qualquer compósito aderido. Não utilize ferramentas que danifiquem a lente.
2) Os medidores de luz diferem muito e são concebidos para pontas e lentes de guia de luz específicas. A Ultradent recomenda verificar rotineiramente a saída no modo Standard
Power. NOTA: a saída numérica verdadeira será distorcida devido à imprecisão dos medidores de luz comuns e do pacote de LED personalizado na luz de polimerização
Manutenção da bateria
Carregamento e troca de pilhas
A luz de polimerização vem com 4 baterias recarregáveis de fosfato de ferro e lítio. A Ultradent recomenda apenas a utilização de pilhas recarregáveis VALO.
Como carregar pilhas:1) Coloque o carregador e pilhas numa área onde não entrarão em contacto com líquidos ou spray desinfetante.
2) Insira as pilhas no carregador com a extremidade positiva (+) apontada para as luzes indicadoras do carregador.
3) As luzes verdes indicam que as pilhas estão prontas para utilização.
4) As pilhas vão demorar de 1 a 3 horas para carregar. Deixe as pilhas no carregador até que estejam prontas para utilização.
NOTA: Se a luz vermelha do carregador não ficar verde quando as pilhas estiverem a carregar por mais de três horas, a bateria pode estar degradada (corrompida) e não pode sercarregada. Experimente uma nova pilha ou ligue para o Serviço de Atendimento ao Cliente Ultradent para solicitar um novo conjunto de pilhas recarregáveis.

Botões TempoOpções de tempo

5sPara alterar o Tempo

10s

15s

20s

1s

2s

3s

4s

Apenas 3sPressione e solte o botão Tempo rapidamente para percorrer as opções de tempo.

Para mudar de modo

Pressione o botão Tempo durante 2 segundos e solte. O fotopolimerizador mudará para o modo seguinte.

LegendaLEDs acesos

LEDs piscando

Pilhas Fracas: A luz de polimerização sinaliza ao utlizador que é hora de trocar as Pilhas quando a Luz Indicadora de Bateria Fraca estiver a piscar em vermelho. Se a carga da bateria
ficar muito baixa, ocorrerá um som sonoro de aviso sonoro de 3 bipes e a luz de polimerização não permitirá operação adicional até que as pilhas sejam recarregadas ou novas pilhas
sejam inseridas. (Consulte o Guia de Aviso Rápido)

Guia Rápido de Polimerização
Modo de polimerização

Potência* (mW)

Como trocar/inserir pilhas:
1) Remova a tampa traseira rodando um quarto de volta. no sentido contrário ao ponteiro dos relógio.
2) Remova as pilhas.
3) Insira pilhas novas com o lado positivo (+) primeiro.
4) Volte a colocar a tampa traseira alinhando e empurrando suavemente no sentido do ponteiro do relógio. A tampa vai fazer um clique quando estiver totalmente encaixada.
5) A unidade está pronta para utilização.

Irradiância* (mW/cm2)

Tempo total de exposição (segundos)

Energia** (Joules)

Standard

970

900

5/10/15/20

4.8J/9.7J/14.5J/19.4J

High Power Plus

1,620

1,500

1/2/3/4

1.6J/3.2J/4.8J/6.5J

Xtra

2,260

2,100

3

6,8J

*Saída nominal a uma distância de 2 mm da ponta da lente até a superfície superior do compósito.
**Os valores na secção de energia total por ciclo (joules) são arredondados para o décimo mais próximo.

Guia Rápido de Advertências:

Tempo de carregamento e vida útil da bateria: A duração da bateria totalmente carregada na luz de polimerização depende do intervalo Modo/Tempo, do tipo de bateria, da quantidadede utilização e da eficiência do LED.
• Intervalo de recarga recomendado: Recarregue conforme necessário ou, quando o indicador de bateria fraca acender. A expectativa de vida depende da utilização.
• Pilhas Extra: A luz de polimerização vem com 4 pilhas recarregáveis.
Reparação realizada pelo fabricante1) As reparações só devem ser realizadas por pessoal de serviço autorizado. A Ultradent deve fornecer ao pessoal de serviço documentação para realizar reparações.
2) Ao enviar unidades para reparação, manutenção ou calibrações, remova sempre as pilhas da luz de polimerização e do carregador. Embrulhe as pilhas, o carregador, o adaptador e a
luz de polimerização separadamente na caixa de devolução.
3) Envie as pilhas de acordo com as regulamentações locais.
Garantia("Ultradent") garante que este produto deverá, por um período de 5 anos a partir da data de compra, quando operado de acordo com as instruções de operação incluídas com oproduto, (i) estar em conformidade em todos os aspectos materiais com as especificações estabelecidas na documentação da Ultradent que acompanha o produto; e (ii) estar livre dedefeitos de material e de fabrico.

Aviso do Nível de Potência

Aviso de Temperatura

Aviso de Serviço

Aviso de LED

Substitua as baterias

Deixe esfriar

Ligue para a Assistência ao clientepara reparo

Ligue para a Assistência ao clientepara reparo

Esta garantia limitada é intransferível e aplica-se exclusivamente ao comprador original e não se estende aos proprietários subsequentes do produto. Esta garantia limitada não cobrequaisquer outros componentes acessórios, tais como, mas não limitado a, pilhas, carregadores, adaptadores ou lentes adaptáveis. Esta garantia limitada é nula se o produto falhar ou
for danificado devido a negligência, abuso, má utilização, acidente, modificação, adulteração, alteração ou falha em seguir as instruções de utilização aplicáveis. Apenas para fins deexemplo, um produto que é sofre uma queda e é danificado não é coberto por esta garantia. Para se qualificar sob esta garantia limitada, o comprovativo de compra (por exemplo,
recibo de venda ou documentação similar) deve ser apresentado à Ultradent juntamente com o produto defeituoso.

• Sem som
• A piscar, todos os 2 segundos• Permite a operação

• 3 bips contínuos
• Flash rápido simples• Proíbe as operações

Um produto defeituoso que atenda às condições de garantia aqui estabelecidas será, a critério exclusivo da Ultradent, reparado ou substituído. Em nenhuma hipótese a
responsabilidade da Ultradent pelo produto excederá o preço de compra pago pelo comprador. Sob nenhuma circunstância a Ultradent será responsável por quaisquer danos indiretos,
incidentais, previstos, imprevistos, especiais ou consequenciais decorrentes de ou em ligação com a utilização deste produto.

• Bateria fraca: piscar devagar• Desligar: 3 bips, piscando• Proíbe a operação

• 3 bips• Piscar lento• Proíbe a operação

Processamento

Este produto destina-se a ser utilizado com uma manga barreira. Veja acima para manutenção e limpeza geral.

Armazenamento e Descarte

Se guardar o fotopolimerizador por períodos superiores a 2 semanas, ou se o embalar para uma viagem, remova sempre as baterias. Se as baterias forem deixadas na unidade por
longos períodos de tempo sem recarga podem tornar-se não funcionais ou não-carregáveis. NÃO guarde as baterias a temperaturas acima de 40°C (104ºF) ou sob luz solar direta.

Manutenção

Limpeza Geral da Luz de Polimerização
Após cada utilização, humedeça uma gaze ou pano macio com um desinfetante de superfície aprovado e limpe a superfície e a lente. Produtos de limpeza não autorizados podem
causar danos à luz de polimerização.
PRODUTOS DE LIMPEZA ACEITÁVEIS:• Álcool isopropílico a 70%• Etanol a 70%

Limpeza Geral do Escudo de Luz:
Limpe o escudo de luz sem fios VALO utilizando qualquer desinfectante de superfície. NÃO autoclave.
Reparação
Manutenção realizada pelo utilizador
64

Armazenamento e transporte do fotopolimerizador:• Temperatura: +10°C a +40°C (+50°F a +104°F)• Umidade relativa: 10% a 95%• Pressão ambiente: 500 hPa a 1060 hPa
Ao descartar resíduos eletrônicos (por exemplo, fotopolimerizadores, carregadores, baterias e fontes de alimentação), siga as diretrizes locais para resíduos e reciclagem.

Considerações Técnicas
Acessórios

65

Item

Informação da CE

Atributo

Informação/Especificação

Protetor de luz VALO Sem fio

Fotopoli-merizador
VALO GrandeClassificações: IEC 60601-1 (Segurança), IEC
60601-1-2 (EMC)

Peso:• Com baterias: 6 oz. (190 gramas)• Sem baterias: 5 oz. (150 gramas)
Dimensões: (8 x 1,28 x 1,06) polegadas, (203 x 32,5 x 27) mm

Carregador VALO

Fonte dealimentação docarregador

Saída - 12VCC a 500mA
Entrada – 100VCA a 240VCA
Ultradent P/N 5930 Fonte de alimentação do
Carregador VALO com acessórios de plugueinternacional

Classificações: IEC 60601-1 (Segurança)
Comprimento do cabo - 6 pés (1,8 metros)
A fonte de alimentação do carregador VALO é uma fonte de alimentação Classe
II de classe médica e fornece isolamento da rede elétrica

Carregador VALO

Carregador de baterias inteligente VALO para baterias de fosfato de ferro-lítio de 3,6VCC:• Desligamento automático quando totalmente carregada• Detecção automática de baterias com defeito• Proteções: Sobrecarga térmica, curto-circuito, polaridade inversa
⸰ LED vermelho - Carregamento⸰ LED verde - Descarregada ou completamente carregada
⸰ LED apagado – curto-circuito• O tempo de carga: 1 - 3 horas
Qualificação: CE, WEEE

Baterias VALO

Recarregáveis: Química segura Fosfato de ferro de lítio (LiFePO4) RCR123A• Tensão de trabalho: 3,2VCC
Classificações: UL, CE, RoHS, WEEE

Condições defuncionamento

Temperatura: +10°C a +32°C (+ 50°F a + 90°F)
Umidade relativa: 10% a 95%
Pressão ambiente: 700 hPa a 1060 hPa

Ciclo de trabalho

O fotopolimerizador foi projetado para operações de curta duração. À temperatura ambiente máxima (32°C) 1 minuto de ciclo
Ligado, 30 minutos desligado (período de arrefecimento).

Baterias VALO

Informação técnica/Dados

Bandas de comprimentos de onda efetivas para polimerização de compósitos:

BANDAS DE COMPRIMENTO DE ONDA DE CURA COMPÓSITA EFICAZES(Faixa de comprimento de onda utilizável: 385-515 nm)POTÊNCIA RADIANTE ESPECTRAL (mW/nm)

Efetivo inferior
banda de comprimento de onda
395-415 NM

Banda decomprimento de onda
superior efetiva
440-480 nm

XtraHigh PowerStandard

Resolução de problemas
Se as soluções sugeridas abaixo não corrigirem o problema, ligue para a Ultradent através do número 800.552.5512. Fora dos Estados Unidos, contacte o seu
distribuidor Ultradent ou revendedor de material dentário.
Problema

COMPRIMENTO DE ONDA (nm)Atributo

Informação/Especificação

Lente

Diâmetro 11,7 mm

Intervalo decomprimento
de onda• Intervalo de comprimento de onda utilizável: 385 – 515nm• Comprimentos de onda de pico: 395 – 415nm e 440 – 480nm
Tabela de Comparação de Exitância Radiante Nominal
Instrumento demedição

Tabela deIntensidade de Luz

66‡ Analisador de espectro Gigahertz
Exitância

Potência total

Abertura do
medidor

15 mm

15 mm

Potência Standard(±10%)

900 mW/cm²

970 mW

Alta Potência Plus(±10%)

1500 mW/cm²

1615 mW

Potência Xtra(±10%)

2100 mW/cm²

2260 mW

A Saída Radiante varia de acordo com acapacidade do instrumento, o método demedição e o posicionamento da luz.
‡ A Saída Radiante está em conformidade
com a ISO 10650 quando medida com um
analisador de espectro Gigahertz-Optik.

Possíveis soluções

A luz não liga

1) Pressione o botão Mudar Tempo/Modo ou o botão de ativação para sair do modo Poupança de energia.
2) Verifique o indicador vermelho de bateria fraca para ver o estado da carga da bateria.
3) Verifique se as baterias novas estão inseridas corretamente na unidade.
4) Se os LEDs de aviso vermelho e amarelo estiverem piscando, isso significa que o fotopolimerizador atingiu seu limite
de segurança de temperatura interna. Deixe o fotopolimerizador esfriar por 10 minutos ou use uma toalha úmida e fria
para esfriar a unidade rapidamente.
5) Se o LED vermelho de Aviso piscar e emitir um bipe continuamente, ligue para o Serviço de Atendimento ao Cliente
Ultradent para reparo.

A luz não fica ligada durante o tempo
pretendido

1) Verifique as luzes modo e de temporização confirmar que tem a temporização correta.
2) Verifique o estado de carga da bateria no indicador de bateria fraca.
3) Verifique se as baterias novas estão corretamente inseridas na unidade.

A luz não polimeriza as resinas
adequadamente

1) Verificar se a lente apresenta resíduos de resinas/compósitos polimerizados.
2) Usando a proteção UV adequada dos olhos, verifique se as luzes LED estão funcionando.
3) Verificar o nível de potência com um fotómetro. Se estiver utilizando um fotómetro, a Ultradent recomenda a verificação
do fotopolimerizador no modo Potência Standard.
NOTA: A saída numérica real será distorcida devido à inexatidão dos fotómetros comuns e ao conjunto LED personalizado
usado pelo fotopolimerizador. Os fotómetros diferem bastante e estão concebidos para pontas de guias de luz e lentes
específicas.
4) Verificar a data de validade da resina de polimerização.
5) Certifique-se de que é seguida a técnica adequada de acordo com as recomendações do fabricante.

As baterias não carregam

1) Assegure-se de que as baterias estão inseridas no carregador com a orientação correta e que as baterias sejam
carregadas durante 1-3 horas.
2) Se as luzes vermelhas no carregador não mudarem para verde, contate o Serviço ao Cliente da Ultradent paraencomendar baterias de substituição e/ou um carregador.
3) Se nem as luzes verdes nem vermelhas do carregador estiverem visíveis, ligue para o Serviço ao Cliente da Ultradentpara encomendar ou substituir o carregador e/ou o adaptador CA.
67

Se as soluções sugeridas abaixo não corrigirem o problema, ligue para a Ultradent através do número 800.552.5512. Fora dos Estados Unidos, contacte o seu
distribuidor Ultradent ou revendedor de material dentário.
ProblemaPossíveis soluções

O carregador não carrega as baterias

1) Certifique-se de que o carregador está ligado e de que o adaptador CA está ligado a uma tomada eléctrica.
2) Se as luzes verdes ou vermelhas no carregador não estiverem visíveis, contate o Serviço ao Cliente da Ultradent para
obter um novo carregador e/ou adaptador AC.

Não é possível alterar o modo ou
intervalos de tempo

1) Mantenha os botões Tempo/Modo e Potência pressionados até que uma série de bipes indique que o fotopolimerizador
está desbloqueado.

Informações diversas
Orientações e declaração do fabricante sobre emissões eletromagnéticas
O fotopolimerizador destina-se a ser usado no ambiente eletromagnético especificado abaixo. O cliente ou usuário deve se certificar de que este será usado nesse
ambiente.
Testes de emissões

Conformidade

Ambiente eletromagnético - orientações

Emissões de RF
CISPR 11

Grupo 1

O fotopolimerizador usa baterias e não é afetado por EMI, RF ou supressão de picos.

Emissões de RF
CISPR 11Classe B

O fotopolimerizador utiliza energia elétrica e eletromagnética somente para suas funções internas.
Portanto, quaisquer emissões de RF são muito baixas e não deverão causar interferência nos
equipamentos eletrônicos mais próximos.

Emissões harmônicas
IEC 61000-3-2

N/A

Os testes de emissão de harmônicos e oscilação de tensão não são aplicáveis ao
fotopolimerizadorss porque ele é alimentado por bateria.

Flutuações de tensão/emissões
intermitentes IEC 61000-3-3

N/A

O fotopolimerizador é adequado para uso em todos os estabelecimentos, inclusive em ambientes
domésticos e em estabelecimentos ligados diretamente à rede de alimentação elétrica pública debaixa tensão, que alimenta os prédios residenciais.

Orientações e declaração do fabricante sobre imunidade eletromagnética
O fotopolimerizador destina-se a ser usado no ambiente eletromagnético especificado abaixo. O cliente ou usuário deve se certificar de que este será usado nesse
ambiente.
Teste de IMUNIDADE

Nível de teste IEC 60601

Nível de conformidade

Orientação sobre ambiente eletromagnético

Descarga eletrostática (ESD)

Contato de ± 8 kV
Ar de ± 15 kV

Contato de ± 8 kV
Ar de ± 15 kV

O ambiente físico deve ser limitado às seguintes
circunstâncias:
1) Código IP: IP202) Não mergulhe em nenhum líquido.
3) Não utilize perto de gases inflamáveis A unidadeé não-APG e não-AP.
4) Limites de umidade para armazenamento:
10% - 95%
5) Limites de temperatura para armazenamento:
10°C - 40°C

Transientes elétricos rápidos/
rajada
IEC 61000-4-4± 2 kV para linhas de alimentação
elétrica± 1 kV para linhas de entrada/saída± 2 kV para linhas de alimentação
elétrica
Nota 1: O fotopolimerizador não
tem portas de E/S

O fotopolimerizador é alimentado por baterias e nãopode ser ligado à rede de alimentação elétrica.

Pico
IEC 61000-4-5± 1 kV entre linhas± 2 kV entre a linha e a terra± 1 kV entre linhas± 2 kV entre a linha e a terra

Como o fotopolimerizador é alimentado por baterias,
não está sujeito a transientes elétricos, picos ou
baixas de tensão, curto-circuitos, interrupções ou
variações da rede elétrica.

Quedas de tensão, curtoscircuitos, interrupções e variações
nas linhas de entrada daalimentação elétrica<5% U(>95% de queda em U durante
0,5 ciclo)<5% U(>95% de queda em U durante
0,5 ciclo)

O carregador de baterias auxiliar pode estar sujeito aessas situações, mas é uma unidade separada e não
crítica para o funcionamento do fotopolimerizador
VALO Sem fio.

40% U(>60% de queda em U durante
5 ciclos)

40% U(>60% de queda em U durante
5 ciclos)

70% U(30% de queda em U durante
25 ciclos)

70% U(30% de queda em U durante
25 ciclos)

IEC 61000-4-2

IEC 61000-4-11

<5% U
<5% U
(>95% de queda em U durante 5 s) (>95% de queda em U durante 5 s)
Nota 2: Auto-recuperações
Frequência da alimentação elétrica 30 A/m
Campo magnético (50/60 Hz)

30 A/mIEC 61000-4-8

Se a tensão da bateria do fotopolimerizador cairpara 4VDC, a unidade não permitirá a operação. O
fotopolimerizador VALO Sem fio desliga-se. Quando
forem instaladas baterias novas e os níveis de tensão
adequados forem restabelecidos, o fotopolimerizador
reiniciará e voltará ao mesmo estado antes da perda
de energia. O fotopolimerizador faz uma autorecuperação em caso de perda de energia.

Os campos magnéticos de frequência da alimentação
elétrica devem estar nos níveis característicos de um
local típico em ambiente residencial, de assistência desaúde doméstica, comercial, hospital ou militar típicos.

Nota 1: O fotopolimerizador não está equipado com portas ou linhas de E/S acessíveis.
68

69

Orientações e declaração do fabricante sobre imunidade eletromagnética para sistemas que não são de suporte à vida
O fotopolimerizador destina-se a ser usado no ambiente eletromagnético especificado abaixo. O cliente ou usuário deve se certificar de que este será usado nesse
ambiente.
Teste de IMUNIDADE

Nível de teste IEC
60601

Nível de conformidade Orientação sobre ambiente eletromagnético

Condução de RF

3 Vrms3 VrmsIEC 61000-4-6

150 kHz a 80 MHz

150 kHz a 80 MHz

Equipamentos de comunicação de RF portáteis e móveis não devem ser usados mais perto
de qualquer parte do fotopolimerizador, incluindo cabos, do que a distância de separação
recomendada, calculada a partir da equação aplicável à frequência do transmissor.
Distância de separação recomendada:

80 MHz a 800 MHz

800 MHz a 2,5 GHz
RF irradiada

3 V/m

3 V/m

Em que P é a potência de saída máxima do transmissor em watt (W) especificada pelo
fabricante do transmissor e d é a distância de separação recomendada em metros (m).
A intensidade de campo de transmissores de RF fixos conforme determinado por uma
avaliação eletromagnética locala, deve ser menor que o nívelb de conformidade em cadafrequência.

IEC 61000-4-3

80 MHz a 2,5 GHz

80 MHz a 2,5 GHz

Podem ocorrer interferências na proximidade de equipamentos marcados com o seguinte
símbolo:

Orientação e Declaração do Fabricante para as distâncias de separação recomendadas entre equipamentos de comunicações por RF portáteis e móveis e o
fotopolimerizador VALO Sem fio
O fotopolimerizador ao uso em um ambiente eletromagnético no qual as perturbações de RF irradiadas são controladas. O usuário do fotopolimerizador podeajudar a evitar interferência eletromagnética mantendo uma distância mínima entre equipamentos de comunicação de RF portáteis e móveis (transmissores) e o
VALO conforme recomendado abaixo, de acordo com a potência máxima de saída do equipamento de comunicação.
Distância de separação de acordo com a frequência do transmissor(metros)Potência de saída máxima nominal
do transmissor(P em Watts)

150 kHz – 80 MHz

80 MHz – 800 MHz

800 MHz – 2,5 GHz

0,01

0,12 metros

0,035 metros

0,07 metros

0,1

0,37 metros

0,11 metros

0,22 metros

1

1,7 metros

0,35 metros

0,7 metros

10

3,7 metros

1,11 metros

2,22 metros

100

11,7 metros

3,5 metros

7,0 metros

O fotopolimerizador foi testado de acordo com a IEC 60601-1-2: 2014 e foi aprovado para intensidades de campo irradiadas de 10 V/m entre 80 MHz a 2,5 GHz. O
valor de 3Vrms corresponde a V1 e o valor 10 V/m corresponde a E1 nas fórmulas acima.
Para os transmissores com potência máxima de saída não listada acima, a distância de separação recomendada em metros (m) pode ser estimada utilizando aequação aplicável à frequência do transmissor, em que P é a potência máxima de saída do transmissor em watts (W) de acordo com o fabricante do transmissor.
NOTA 1 A 80 MHz e 800 MHz, aplica-se a distância de separação para a faixa de frequência mais elevada.
NOTA 2 Estas diretrizes podem não se aplicar em todas as situações. A propagação eletromagnética é afetada pela absorção e reflexão de estruturas, objetos e
pessoas.

NOTA 1 A 80 MHz e 800 MHz, é aplicável a faixa de frequência mais alta.
NOTA 2 Estas diretrizes podem não se aplicar em todas as situações. A propagação eletromagnética é afetada pela absorção e reflexão de estruturas, objetos e
pessoas.
a
As intensidades de campo de transmissores fixos, como as estações-base de rádio (telemóvel/sem fio), telefones e outros rádios terrestres móveis, rádio
amador, transmissão de rádio AM e FM, e transmissão de TV não podem ser teoricamente previstas com precisão. Para avaliar o ambiente eletromagnético
devido aos transmissores de RF fixos, deve ser considerado um levantamento eletromagnético dos locais. Se a intensidade de campo medida no local em queo fotopolimerizador é usado exceder o nível de conformidade de RF aplicável acima, o fotopolimerizador VALO Sem fio deve ser observado para confirmar oseu funcionamento normal. Se for observado um desempenho anormal, podem ser necessárias medidas adicionais, como a reorientação ou o deslocação do
fotopolimerizador para outro lugar.
b
Na gama de frequências de 150 kHz a 80 MHz, as intensidades de campo devem ser inferiores a 3 V/m.

70

71

VALO™ GRAND

SVTRÅDLÖST HÄRDNINGSLJUS
Produktbeskrivning

Med sitt bredbandiga spektrum är VALO Grand härdljuslampa utformad för att polymerisera alla ljushärdande produkter inom våglängdsområdet 385-515 nm enligt ISO 10650. VALOGrand härdljuslampa använder Ultradent VALO uppladdningsbara batterier och batteriladdare. Härdljuslampan är utformad för att sitta i ett standardfäste på den dentala uniten eller kan
specialmonteras med hjälp av fästet som medföljer satsen.

VALO Grand Produktkomponenter:
•
•
•
•
•
•

För att undvika att skada utrustningen, för INTE in fingrar, instrument eller andra föremål i härdljuslampan batterifack.
För att undvika skador på utrustningen, försök INTE att rengöra de guldpläterade kontakterna eller någon del av batterifacket. Ring Ultradents kundtjänst vid eventuella frågor.
För att undvika risken för korskontaminering är hygienskydd avsedda för engångsanvändning.
För att minska risken för korrosion, ta bort hygienskyddet efter användning.
För att minska risken för underhärdade resiner, använd inte härdljuslampan om linsen är skadad.

Stegvisa instruktioner
Förberedelse

1) Ladda batterierna innan du använder härdljuslampan (se avsnittet Batteriunderhåll).
2) Placera ett nytt hygienskydd över härdljuslampan före varje användning och minimera rynkorna över linsen för bästa resultat.

Notera:1 – VALO Grand härdljuslampa
4 – Ultradent VALO uppladdningsbara batterier
1 – Ultradent VALO batteriladdare med medicinsk kvalitet 12VDC nätadapter
1 – VALO Hygienskydd provpaket
1 – VALO Grand ljusskydd
1 - Fäste för ytmontering av härdljuslampa med dubbelhäftande tejpÖversikt över hanteringar:

•
•
•
•
•

• För att förhindra korskontaminering och hjälpa till att förhindra att dentalt kompositmaterial fastnar på linsens yta och handstycket, måste ett Ultradent-godkänt hygienskydd
användas över VALO härdljuslampa vid varje användning. Hygienskydd är avsedda för engångsanvändning.

VALO Trådlös Ljussköld:

VALO trådlösa ljussköld är ovalformad, kan roteras för maximal användning och kan användas med en transparent skyddsfolie.

Användning

Foto

Knappar och Ljus

Förklaring

Strömbrytare(Topp eller botten)
Tidsinställningslampor
Varningslampa för urladdat batteri
Mode/Status (Läge/Status)-lampa
Tids-/lägesändringsknapp

1) Varje effektläge används för härdning av tandvårds-material med fotoinitiatorer. Se Snabblägesguide för rekommenderade härdningstider.
NOTERA: Härdlampan är programmerad för att flytta från Standardeffekt via Hög effekt Plus till Xtra effekt-läget i ordningsföljd. Om du t.ex. vill byta från Standardeffekt- till Xtra effektläget, måste du gå in i Hög effekt Plus-läget och sedan till Xtra effekt-läget.
2) Härdningsljuset lagrar det senast använda tidsintervallet och läget, och det kommer att återgå till detta när lägena ändras eller om batterierna tas bort.

DriftHÄRDNINGSSLÄGE: Standard strömläge

TIDSINSTÄLLNINGSINTERVALLER: 5, 10, 15, och 20 sekunder.
• Härdningsljuset går automatiskt till det här läget när den FÖRSTA gången sätts på. Läge/status-lampan blir grön och de fyra gröna Tidsinställningslamporna lyser, vilket innebär
Standardeffekt-läge.
• För att ändra tidsintervall trycker man snabbt på tid/läge-knappen.
• För att härda, tryck på strömbrytaren. För att sluta härda innan ett tidsintervall har slutförts, tryck på strömbrytaren igen.

HÄRDNINGSSLÄGE: Hög effekt-plusläge

Källan för belysning för härdning av fotoaktiverade restorativa och självhäftande material.

TIDSINSTÄLLNINGSINTERVALLER: 1, 2, 3, och 4 sekunder.
• Från standard strömläge, tryck och håll in knappen för ändring av tid/läge i 2 sekunder och släpp sedan. läge/status-lampan blir orange och de fyra gröna Tidsinställningslampornalyser och blinkar, vilket innebär Hög effekt-läge.
• För att ändra tidsintervall trycker man snabbt på tid/läge-knappen.
• Tryck på någon Strömbrytaren för att härda. För att avbryta härdningen innan ett tidsintervall har slutförts, tryck på en av Power-knapparna igen.
• För att återgå till standard strömläge, tryck och håll in knappen för ändring av tid/läge i 2 sekunder och släpp, detta växlar till Xtra effekt-läget. Tryck och håll nere i 2 sekunder och
släpp. läge/status-lampan blir grön och de fyra gröna Tidsinställningslamporna lyser, vilket innebär standardläge.

Varningar och försiktighetsåtgärder

HÄRDNINGSSLÄGE: Xtra effektläge(Botten)(Topp)

För alla produkter som beskrivs här: läs noggrant och sätt dig in i alla instruktioner och SDS-information före användning.

Indikationer för användning / avsett syfte

Riskgrupp 2
VARNING Denna produkt emitterar UV. Ögon eller hudirritation kan uppstå vid exponering. Använd lämplig strålskärmning.
VARNING Farlig optisk strålning kan utgå från denna produkt. Se inte direkt in i härdningslampan. Kan vara skadligt för ögonen.
• Titta INTE direkt in i ljusutgången. Patient, tandläkare och assistenter ska alltid bära bärnstensfärgade UV-ögonskydd när härdljuslampan används.
• För att förhindra risken för elektriska stötar är ingen modifiering av denna utrustning tillåten. Använd endast de medföljande Ultradent VALO-strömförsörjnings- ochkontaktadaptrarna. Om dessa komponenter är skadade, använd inte och ring Ultradents kundtjänst för att beställa en ersättning.
• Bärbar RF-kommunikationsutrustning kan försämra prestandan om den används närmare än 30 cm (12 tum).
• Använd endast godkända tillbehör, kablar och nätaggregat för att förhindra felaktig användning, ökad elektromagnetisk strålning eller minskad elektromagnetisk immunitet (seavsnittet Elektromagnetisk strålning).
• För att undvika risken för elektrisk brand i samband med hantering av batterier:⸰ Blanda INTE uppladdningsbara batterier med icke-uppladdningsbara batterier eller andra batterityper.
⸰ FÖRSÖK INTE att ladda icke-uppladdningsbara batterier. Att ladda icke uppladdningsbara batterier kan resultera i person- eller egendomsskador.
⸰ Autoklavera eller spraya INTE batterier, batterikontakter, laddare eller nätadapter med vätska av något slag. Om korrosion uppträder på batteriladdarens kontakter, ring
Ultradents kundtjänst för att beställa en ersättning.
⸰ Ladda INTE batterier i närheten av brandfarliga material.
⸰ Förvara INTE laddaren i behandlingsrummet.
• För att undvika risken för skador, ANVÄND INTE batterier som är korroderade (rost), buckliga, avger en lukt eller vätskor, har ett trasigt eller saknat omslag, eller som på annat sätt ärskadade. Ring Ultradents kundtjänst för att beställa ersättningsbatterier.
• Använd endast rekommenderade batterier. Alternativa batterier kan orsaka felfunktion.
• För att undvika risken för termisk irritation eller skada, undvik att använda härdningscykler direkt efter varandra och exponera inte oral mjukvävnad på nära håll i mer än 10 sekunderi något läge. Om längre härdningstider krävs, använd flera härdningscykler med viloperioder mellan cyklerna, eller använd en dualhärdande produkt för att undvika uppvärmning av
mjukvävnad.
• Var försiktig vid behandling av patienter som lider av ogynnsamma fotobiologiska reaktioner eller känslighet, patienter som genomgår kemoterapibehandling eller patienter som
behandlas med fotosensibiliserande medicin.
• Denna enhet kan vara känslig för starka magnetiska eller statiska elektriska fält, vilket kan störa programmeringen. Om du misstänker att detta har inträffat, koppla ur enheten
tillfälligt och anslut den sedan till uttaget igen.
• Torka INTE av härdljuslampan med frätande eller slipande rengöringsmedel, autoklavera INTE eller doppa ner i någon form av ultraljudsbad, desinfektionsmedel, rengöringslösningeller vätska. Underlåtenhet att följa medföljande bearbetningsinstruktioner kan göra att härdljuslampan inte fungerar.
72

TIDSINSTÄLLNINGSINTERVALLER: Endast 3 sekunder (Notera: Xtra effekt-läget har en 2 sekunders säkerhetsfördröjning vid slutet av varje härdningsperiod för att begränsa
uppvärmning under konsekutiv härdning. Vid slutet av fördröjningen indikerar en ljudsignal att enheten är redo för fortsatt användning).
• Från standard strömläge, tryck på knappen för byte av tid/läge i 2 sekunder, släpp, håll nedtryckt i 2 sekunder och släpp. Läge/status-lampan blir orange och blinkar och tre av degröna Tidsinställningslamporna tänds och blinkar vilket innebär Xtra effekt-läget.
• För att härda, tryck på strömbrytaren. För att avsluta härdningen före tidsintervallet fullbordats, tryck på nytt på strömbrytaren.
• För att återgå till standard strömläge, tryck och håll nere tid/läge-knappen i 2 sekunder och släpp. Läge/status-lampan blir grön och de gröna Tidsinställningslamporna lyser, vilket
innebär Standardeffekt-läge.

Viloläge: Härdlampan går in i viloläge efter 60 sekunder av inaktivitet, vilket indikeras av en långsam blinkning av läge/status-lampan. När du hämtar upp eller rör på enheten kommer
härdningsljuset att “vakna” och systemet återställs automatiskt till den senast använda inställningen. För att maximera batterilivslängden, lämna härdningsljuset ostört när det inteanvänds.

Rengöring1) Kassera använda skyddsfolier i standardavfallet efter varje patient.
2) Se avsnittet Behandling.

Instruktioner för placering av fäste

1) Fästet ska placeras på en plan, oljefri yta.
2) Rengör ytan med tvättsprit.
3) Ta bort baksidan på tejpen till fästet.
4) Placera fästet så att härdningsljuset lyfts uppåt när det tas bort. Tryck det på plats ordentligt.

73

Snabb-läges-Guide:
Standardeffekt
1000 mW/cm2

Läge

Hög effekt Plus
1600 mW/cm2

Xtra effekt
3200 mW/cm2

Strömbrytare

Hur man laddar batterier:
1) Placera laddare och batterier på ett område där de inte kommer i kontakt med vätskor eller desinfektionsspray.
2) Sätt i batterierna i laddaren med den positiva (+) änden riktad mot indikatorlamporna på laddaren.
3) Gröna lampor indikerar att batterierna är klara att användas.
4) Batterierna tar 1-3 timmar att ladda. Lämna batterierna i laddaren tills de ska användas.
OBS: Om den röda lampan på laddaren inte lyser grönt när batterierna har laddats längre än tre timmar, kan batteriet vara degraderat (skadat) och kan inte laddas. Prova ett nytt batteri
eller ring Ultradent kundtjänst för att beställa en ny uppsättning uppladdningsbara batterier.

Läge/Timing-Ledljus

Tidinställningsknappar

Tidsalternativ

5s

10s

15s

20s

1s

2s

3s

4s

Endast 3sÄndra tiden

Tryck och släpp Tidinställningsknappen snabbt för att växla genom tidsalternativen.

Ändra lägen

Tryck och håll ner Tidinställningsknappen i 2 sekunder och släpp den. Härdningsljuset kommer att växla till nästa läge.

Förklaring

Solida Ledljus

Blinkande Ledljus

Snabbguide för Härdning:
Härdningsläge

Effekt* (mW)

Irradians* (mW/cm2)

Total exponeringstid (sekunder)

Energi** (Joule)Standard

970

900

5/10/15/20

4,8J/9,7J/14,5J/19,4J

High Power Plus

1 620

1 500

1/2/3/4

1,6J/3,2J/4,8J/6,5J

Xtra

2 260

2 100

3

6,8J

Snabbvarningsguide:

2) Ljusmätare skiljer sig mycket åt och är designade för specifika ljusledarspetsar och linser. Ultradent rekommenderar att man rutinmässigt kontrollerar effekten i läget Standard Power.
OBS: det verkliga numeriska resultatet kommer inte att vara exakt på grund av låg nogrannhet hos vanliga ljusmätare och det speciella LED-paketet i härdljuslampan
Batteriunderhåll
Ladda och byta batterier
Härdljuslampan levereras med 4 uppladdningsbara litiumjärnfosfatbatterier. Ultradent rekommenderar att du endast använder VALO uppladdningsbara batterier.

*Nominell effekt på ett avstånd av 2 mm från linsens spets till kompositens ovansida.
**Värdena i sektionen för total energi per cykel (joule) avrundas uppåt till närmaste tiondel.

Effektnivå-Varning

Temperatur-Varning

Service-Varning

LED-Varning

Byt ut batterierLåt svalna

Ring kundservice för reparation

Ring kundservice för reparation• Lågt batteri: långsamt blinkande• Stänga av: 3 pipsignaler, blinkar• Förbjuder användning• 3 pipsignaler• Långsamt blinkande• Förbjuder användning• Inget ljud
• Blinkar varannan sekund• Tillåter drift• Oavbrutet, 3 pipsignaler• En enda snabb blinkning• Förbud mot verksamhetSå här byter/sätter du i batterier:1) Ta bort bakstycket genom att vrida moturs ett kvarts varv.
2) Ta ur batterierna.
3) Sätt i nya batterier med plussidan (+) först.
4) Sätt tillbaka locket genom att rikta in och försiktigt trycka samtidigt som du vrider medurs. Locket klickar när det är helt påsatt.
5) Enheten är klar att användas.
Låga batterier: Härdljuslampan signalerar användaren att det är dags att byta batterier när indikatorlampan för lågt batteri blinkar rött. Om batteriladdningen blir för låg hörs ett
varningsljud på tre pip och härdljuslampan tillåter inte ytterligare drift förrän batterierna laddas upp eller nya batterier sätts i. (Se Snabbguide för varningar)
Laddningstid och batteritid: Batteritiden för fulladdat batteri i härdljuslampan beror på läget/timerintervallet, batterityp, användningsmängd och LED-effektivitet.
• Rekommenderat laddningsintervall: Ladda vid behov eller när indikatorn för låg batterinivå tänds. Den förväntade livslängden beror på användningen.
• Extra batterier: Härdljuslampan levereras med 4 uppladdningsbara batterier.
Reparation av tillverkaren
1) Reparationer får endast utföras av auktoriserad servicepersonal. Ultradent ska förse servicepersonal med dokumentation för att utföra reparationer.
2) Ta alltid bort batterierna från härdljuslampan och laddaren när du skickar in enheter för reparation, service eller kalibrering. Slå in batterier, laddare, adapter och härdljuslampan
separat i returlådan.
3) Skicka batterier i enlighet med lokala bestämmelser.
Garanti
Ultradent Products, Inc. ("Ultradent") garanterar att denna produkt, under en period av 5 år från inköpsdatumet, när den används enligt de bruksanvisningar som medföljer produkten,(i) i alla väsentliga avseenden överensstämmer med de specifikationer som anges i Ultradents dokumentation som medföljer produkten; och (ii) är fri från defekter i material och
utförande.
Denna begränsade garanti är ej överlåtbar och gäller endast den ursprungliga köparen och sträcker sig inte till efterföljande ägare av produkten. Denna begränsade garanti täcker intenågra andra tillbehörskomponenter som, men inte begränsat till, batterier, laddare, adaptrar eller adaptiva linser. Denna begränsade garanti gäller inte om produkten inte fungerar
eller skadas på grund av försumlighet, missbruk, felaktig användning, olyckshändelse, modifiering, manipulering, ändring eller underlåtenhet att följa tillämpliga bruksanvisningar. Som
exempel kan nämnas att en produkt som tappas och skadas inte täcks av denna garanti. För att omfattas av denna begränsade garanti måste inköpsbevis (t.ex. kvitto eller liknande
dokumentation) skickas till Ultradent tillsammans med den defekta produkten.
En defekt produkt som uppfyller de garantivillkor som anges häri kommer, efter Ultradents eget gottfinnande, antingen att repareras eller bytas ut. Under inga omständigheter ska
Ultradents ansvar för produkten överstiga det inköpspris som betalats av köparen. Under inga omständigheter kan Ultradent hållas ansvarigt för indirekta, tillfälliga, förutsedda,
oförutsedda, speciella eller följdskador som uppstår på grund av eller i samband med användningen av denna produkt.

Behandling

Denna produkt är avsedd att användas med ett hygienskydd. Se ovan för underhåll och allmän rengöring.

Förvaring och Avfallshantering

Ta bort batterierna om härdningsljuset lagras under längre perioder än 2 veckor eller packas för att resa. Om batterierna lämnas i enheten under långa perioder utan laddning kan de bli
funktions- eller laddningsodugliga. Förvara inte batterier i temperaturer över 40 °C (104°F) eller i direkt solljus.
Härdningsljus Förvaring och Transport:• Temperatur: +10 °C till + 40 °C (+ 50 °F till + 104 °F)• Relativ luftfuktighet: 10 % till 95 %• Omgivningstryck: 500 hPa till 1060 hPa

Underhåll

Allmän rengöring av härdljuslampan
Fukta en gasväv eller mjuk trasa med ett godkänt ytdesinfektionsmedel efter varje användning och torka av ytan och linsen. Otillåtna rengöringsmedel kan orsaka skador på
härdljuslampan.
GODTAGBARA RENGÖRINGSMEDEL:• 70 % isopropylalkohol• 70 % etanol

Allmän rengöring av ljusskyddet:
Rengör VALO sladdlösa ljusskydd med valfritt ytdesinfektionsmedel. Autoklavera INTE.
Reparation
Användarutfört underhåll
1) Använd ett hygienskydd för att förhindra att dental komposit fäster på linsens yta. Använd vid behov ett tandinstrument av plast eller rostfritt stål för att försiktigt ta bort eventuellvidhäftad komposit. Använd inte verktyg som skadar linsen.
74

Vid hantering av elektroniskt avfall (d.v.s. härdningsljus, laddare, batterier och strömförsörjning) ska man följa lokala avfalls- och återvinningsriktlinjer.

Tekniska Hänsynstagande
Tillbehör
ArtikelCE-information

VALO Trådlös Ljussköld

75

ArtikelCE-information

Teknisk information / Data

Effektiva komposithärdande våglängdsband:

VALO laddare

EFFEKTIVE KOMPOSITHÆRDENDE BÅLÆNGDEBÅND
(Anvendeligt bølgelængdeområde: 385-515 nm)

SPEKTRAL STRÅLEKRAFT (mW/nm)

VALO Batterier

Effektiv lavere
bølgelængdebånd
395-415 nm

Effektivt øvrebølgelængdebånd
440-480 nm

XtraHigh PowerStandard

WAVEL ENGTH (nm)Egenskaper

Information / specifikation

Lins

Diameter 11,7 mm

Våglängds-intervall• Användbart våglängdsintervall: 385 - 515 nm• Toppvåglängder: 395 - 415nm och 440 - 480nm
Tabell över Nominell Jämförelse av Strålningsutgång
Mätins-trument

Ljusstyrke-tabell76‡ Gigahertz-spektrumanalysatorUtgångTotal-effekt

Mätar-öppning

15 mm

15 mm

Standard-effekt(±10%)

900 mW/cm²

970 mW

Hög effekt Plus(±10%)

1500 mW/cm²

1615 mW

Xtra effekt (±10%)2100 mW/cm²

Strålningsexitansen kommer att variera
beroende på instrumentets kapacitet,mätmetod och ljusplacering.
‡ Strålningsexitansen överensstämmer med
ISO 10650 när den mäts med en GigahertzOptik spektrumanalysator.

2260 mW

VALO Grand
Härdnings-ljus

Klassificeringar: IEC 60601-1 (Säkerhet), IEC 60601-1-2
(EMC)

Vikt:
• Med batterier: 6 oz. (190 gram)• Utan batterier: 5 oz. (150 gram)Mått: (8 x 1,28 x 1,06) tum, (203 x 32,5 x 27) mm

Strömför-sörjning
för LaddareUtgång - 12 VDC vid 500mAIngång – 100 VAC till 240 VAC
Ultradent P / N 5930 VALO Laddare Strömförsörjning
med internationella plugginsatser

Klassificeringar: IEC 60601-1 (Säkerhet)
Ledningslängd - 6 fot (1,8 meter)
Strömförsörjningen till VALO Charger Power Supply (Laddare) är enströmförsörjning i medicinsk Klass II och är isolerad från nätspänningen

VALO laddare

VALO 3.6VDC Litiumjon-fosfat smart batteriladdare:• Automatisk avstängning när den är fulladdat• Automatisk detektering av defekta batterier• Skydd: Termisk, Överladdning, Kortslutning, omvänd polaritet
⸰ Röd LED - Laddar
⸰ Grön LED - Tom eller helt laddad
⸰ LED av - kortslutning• Laddningstid: 1 - 3 timmar
Klassificering: CE, WEEE
77

Diverse Information

Egenskaper

Information / specifikation

VALO Batterier

Laddningsbara: Säkert kemiskt litiumjon-fosfat(LiFePO4) RCR123A• Arbetsspänning: 3,2 VDC
Klassificeringar: UL, CE, RoHS, WEEE

Vägledning och tillverkningsdeklaration för elektromagnetiska emissioner

Driftsför-hållanden

Temperatur: + 10 °C till + 32 °C (+ 50 °F till + 90 °F)
Relativ luftfuktighet: 10 % till 95 %Omgivningstryck: 700 hPa till 1060 hPa

EmissionstestÖverens-stämmelse

Elektromagnetisk miljö – riktlinjer

RF-emissioner
CISPR 11

Grupp 1

Härdningsbelysningen använder batterier och påverkas inte av EMI, RF eller överspänning.

RF-emission
CISPR 11

Klass B

Härdningsbelysningen använder endast elektrisk och elektromagnetisk energi för sina interna
funktioner. Därför är alla RF-emissioner mycket låga och kan inte förorsaka störningar inärliggande elektronisk utrustning.

Harmoniska emissioner
IEC 61000-3-2

Inte tillämpbart

Harmonisk emission och spänningsfluktuationer behöver inte beaktas, då härdningsljuset ärbatteridrivet.

Spänningsfluktuationer/ flimmeremissioner IEC
61000-3-3

Inte tillämpbart

Härdlampan är lämplig för användning i alla anläggningar, inklusive lokala anläggningar och desom är direkt anslutna till det allmänna nätverket för den lågspänningsförsörjning som levereras
till hushåll.

Härdlampan är avsedd att användas i den elektromagnetiska miljön som anges nedan. Kunden eller användaren ska se till att den används i en sådan miljö.

Härdningsljuset är konstruerad för kortvarig drift. Vid maximal omgivningstemperatur (32 °C) 1 minut ON back-to-back period, 30 minuter
AV (kyltid).

Driftperiod

Felsökning
Om lösningarna som föreslås nedan inte löser problemet, vänligen ring Ultradent på +1 800 552 5512. Utanför USA: ring din Ultradent-distributör eller återförsäljare
av tandvårdsmaterial.
ProblemMöjliga lösningar

Ljuset tänds inte

1) Tryck på tid/läge eller ström-knappen för att väcka systemet från strömsparläget.
2) Kontrollera den röda låg-batterinivå-lampan som anger batteriets laddningsnivå.
3) Kontrollera att nya batterier har satts rätt in i enheten.
4) Om de röda och gula Varnings-Ledljusen blinkar, betyder det att härdningsljuset har nått sin interna
temperaturbegränsning. Låt härdningsljuset svalna i 10 minuter eller använd en sval fuktig handduk för att kyla ner
enheten snabbt.
5) Om det röda LED-varnings-Ledljuset oavbrutet blinkar och det piper ska Ultradents kundtjänst kontaktas för reparation.

Ljuset blir inte på för önskad tid1) Kontrollera Läge och Tidsinställningslampor för korrekt tidsinmatning.
2) Kontrollera låg batterinivå-lampan som anger batteriets laddningsnivå.
3) Kontrollera att de nya batterierna har satts rätt in i enheten.

Ljuset härdar inte hartserna ordentligt

1) Kontrollera att det inte finns överbliven härdad harts/komposit på linsen.
2) Använd korrekt UV-skyddsglasögon, kontrollera att LED-ljusen fungerar.
3) Kontrollera effektnivån med en ljusmätare. Om du använder en ljusmätare rekommenderar Ultradent att kontrollera
härdningsljuset i standard strömläge (Standardeffektläge).
NOTERA: Den sanna numeriska utmatningen kommer att vara snedställd på grund av felaktigheten hos vanliga ljusmätare
och det anpassade LED-paketet som härdningsljuset använder. Ljusmätare varierar avsevärt och är utformade för specifika
ljusledningstoppar och linser.
4) Kontrollera bäst-före-datum för härdningshartsen.
5) Se till att korrekt förfaringssätt följs i enlighet med tillverkarens rekommendationer.

Batterierna laddas inte

1) Se till att batterierna sitter i laddaren i rätt läge och låta batterierna ladda i 1-3 timmar.
2) Om röda ljus på laddaren inte växlar till grönt, ring Ultradent kundservice för att beställa bytesbatterier och/eller laddare.
3) Om varken det gröna eller det röda ljuset på laddaren lyser, ring Ultradent kundservice för att beställa eller byta ut
laddare och/eller nätadapter.

Laddaren laddar inte batterier

1) Kontrollera att laddaren är inkopplad och att el-adaptern är instucken i ett fungerande eluttag.
2) Om varken det gröna eller det röda ljuset på laddaren lyser, ring Ultradent Kundtjänst för en ny laddare och/eller
nätadapter.

Går inte att ändra läge eller
tidsintervaller

Riktlinjer och tillverkningsdeklaration för elektromagnetisk immunitet
Härdlampan är avsedd att användas i den elektromagnetiska miljön som anges nedan. Kunden eller användaren ska se till att den används i en sådan miljö.
IMMUNITETS-TEST

IEC 60601 testnivå

Nivå för överensstämmelse

Riktlinjer för Elektromagnetisk omgivning

Elektrostatisk urladdning (ESD)± 8 kV kontakt± 15 kV luft

± 8 kV kontakt± 15 kV luft

Den fysiska omgivningen bör begränsas till följande:
1) IP-beteckning: IP20
2) Sänk inte ner i vätska.
3) Använd inte i närheten av brandfarliga gaser.
Enheten är icke-APG och icke-AP.
4) Område för luftfuktighet vid förvaring: 10% - 95%
5) Temperaturområde vid lagring: 10 °C – 40 °C

Snabba elektriska transienter/”salvor”
IEC 61000-4-4± 2 kV för kraftöverföringsledningar± 1 kV för ingångs-/
utgångsledningar± 2 kV för kraftöverföringsledningar
Anmärkning 1: Härdningsljuset
har inga I/O-portar

Härdningsljuset är batteridrivet och kan inte anslutas
till 230 V-nätet.

Ökning
IEC 61000-4-5± 1 kV ledning till ledning± 2 kV ledning till jord

± 1 kV ledning till ledning± 2 kV ledning till jord

Eftersom härdningsljuset är batteridrivet, är det
inte utsatt för elektriska transienter, störningar,
spänningsfall, kortslutning, avbrott eller variationer
i 230 V-nätet .

Spännings-dippar, kortslutningar
och variationer i ingående
strömför-sörjning<5 % U(> 95 % dip i U under 0,5 period)<5 % U(> 95 % dip i U under 0,5 period)

Batteriladdaren, som är ett tillbehör, kan vara utsatt
för ovanstående men den är galvaniskt isolerad ochinte kritisk för driften av VALO Trådlösa härdningsljus.

40 % U(60 % dip i U under 5 cykler)

40 % U(60 % dip i U under 5 cykler)

70 % U(30 % dip i U under 25 cykler)

70 % U(30 % dip i U under 25 cykler)<5 % U(> 95 % dip i U i 5 s)

<5 % U(> 95 % dip i U i 5 s)Anmärkning 2: Återaktiverar
automatiskt

30 A/m30 A/mIEC 61000-4-2

IEC 61000-4-11

1) Håll inne både tid/läge- och ström)-knapparna tills en serie pip indikerar att härdningsljuset är upplåst.

Frekvens(50/60 Hz) magnetfält
IEC 61000-4-8

Om batterispänningen i härdningsljuset sjunker
till 4 VDC, tillåter enheten inte drift. VALO Trådlösa
härdningsljus stängs av. När nya batterier sätts
in och rätt spänningsnivåer återställs, återupptar
härdningsljuset driften och återgår till samma
tillstånd som innan strömförlusten uppstått.
Härdningsbelysningen utför en självåterhämtning
efter en blackout.
Magnetfält med strömfrekvens bör vara på nivåer somär karakteristiska för en typisk plats i en omgivningsom är typisk för bostäder, hemsjukvård, kontor,
sjukhus eller militär miljö.

Anmärkning 1: Härdningsljuset är inte utrustat med några portar eller några tillgängliga I/O-linjer.

78

79

Handledning och tillverkningsdeklaration för elektromagnetisk immunitet för livförsörjningssystem
Härdlampan är avsedd att användas i den elektromagnetiska miljön som anges nedan. Kunden eller användaren ska se till att den används i en sådan miljö.
IMMUNITETS-TEST

IEC 60601 testnivå

Nivå för överensstämmelse

Lednings-bundna RF

3 Vrms3 VrmsIEC 61000-4-6

150 kHz till 80 MHz

150 kHz till 80 MHz

Utstrålad
radiofrekvens

3 V/m

3 V/m

Riktlinjer för Elektromagnetisk omgivning
Bärbar och mobil RF-kommunikationsutrustning får inte användas närmare någon delav härdningsljuset, inklusive kablar, än det rekommenderade avskärmningsavståndet
beräknat från ekvationen som gäller för sändarens frekvens.
Rekommenderat avskärmningsavstånd:80 MHz till 800 MHz

800 MHz till 2,5 GHz
P är sändarens maximala utgångseffekt i watt (W) enligt sändarens tillverkare och d är det
rekommenderade avskärmningsavståndet i meter (m).
Fältstyrkor från stationära RF-sändare, som upptäcks vid en elektromagnetisk
platsundersökninga, bör vara mindre än överensstämmelsesnivån i varje frekvensområdeb.

IEC 61000-4-3

80 MHz till 2,5 GHz

80 MHz till 2,5 GHzInterferens kan inträffa i närheten av utrustning märkt med följande symbol:

ANM. 1 Vid 80 MHz och 800 MHz gäller det högre frekvensområdet.
ANM. 2 Dessa riktlinjer gäller eventuellt inte i alla situationer. Elektromagnetisk utbredning påverkas av absorption och reflektion från byggnader, föremål och
människor.
aFältstyrka från stationära sändare, såsom basstationer för radio (mobila/trådlösa) telefoner och landbaserade mobilradionät, amatörradio, radiosändningar påAM och FM och TV-sändningar kan inte teoretiskt förutsägas med exakthet. För att fastställa den elektromagnetiska miljö som skapas av stationära RF-sändaremåste en elektromagnetisk platsundersökning övervägas. Om den uppmätta fältstyrkan på den plats där härdningsljuset används överskrider den tillämpliga
RF-överensstämmelsesnivån ovan, bör VALO Trådlösa härdningsljus hållas under observation för att säkerställa normal drift. Om onormal prestanda observeras kan
ytterligare åtgärder vara nödvändiga, såsom omriktning eller omplacering av härdningsljuset.
b
I frekvensområdet 150 kHz till 80 MHz bör fältstyrkorna vara mindre än 3 V / m.

80

Riktnings- och tillverkningsdeklaration för rekommenderade avskärmningsavstånd mellan bärbar och mobil RF-kommunikationsutrustning och VALO Trådlösa
härdningsljus
Härdningsljuset är avsett för användning i en elektromagnetisk omgivning där utstrålade RF-störningar är under kontroll. Användaren av härdningsljuset kan
bidra till att förhindra elektromagnetisk störning genom att hålla ett minimum avstånd mellan bärbar och mobil RF-kommunikationsutrustning (sändare) och
härdningsljuset enligt nedan, baserat på kommunikationsutrustningens maximala uteffekt.
Avskärmningsavstånd beroende på sändarens frekvens(meter)Nominell maximal uteffektfrån sändaren(P i Watt)

150 kHz – 80 MHz

80 MHz – 800 MHz

800 MHz – 2,5 GHz

0,01

0,12 meter

0,035 meter

0,07 meter

0,1

0,37 meter

0,11 meter

0,22 meter

1

1,7 meter

0,35 meter

0,7 meter

10

3,7 meter

1,11 meter

2,22 meter

100

11,7 meter

3,5 meter

7,0 meter

Härdningsljuset har testats enligt IEC 60601-1-2: 2014 och passerat under utstrålade fältstyrkor på 10 V/m i ett område mellan 80 MHz och 2,5 GHz. Värdet på 3Vrms
motsvarar V1 och värdet 10V/m motsvarar E1 i formlerna ovan.
För sändare klassificerade med en maximal utgångseffekt som inte anges ovan kan det rekommenderade avskärmningsavstånd d i meter (m) uppskattas med hjälp
av ekvationen som gäller för sändarens frekvens, där P är sändarens maximala uteffekt i watt (W) enligt sändarens tillverkare.
ANM. 1 Vid 80 MHz och 800 MHz gäller avskärmningsavståndet för det högre frekvensområdet.
ANM. 2 Dessa riktlinjer gäller eventuellt inte i alla situationer. Elektromagnetisk utbredning påverkas av absorption och reflektion från byggnader, föremål och
människor.

81

VALO™ GRAND

DAHÆRDNINGSLYS

Produktbeskrivelse

Med bredbåndsspektret er VALO Stor designet til at polymerisere alle lyshærdede produkter i bølgelængdeområdet 385-515 nm ifølge ISO 10650. VALO Stor hærdningslys bruger
Ultradent VALO genopladelige batterier og batterioplader. Hærdningslyset er beregnet til at hvile i en standardkonsol til dentale enheder eller monteringen kan tilpasses ved hjælp af
den konsol, som følger med sættet.

VALO Stor produktkomponenter:
•
•
•
•
•
•

1 - VALO Stor hærdningslys
4 - Ultradent VALO genopladelige batterier
1 - Ultradent VALO batterioplader med 12 VDC vekselstrømsadapter af medicinsk kvalitet
1 - VALO Barrieremuffe, prøvepakke
1 - VALO Stor Lysskærm
1 - Konsol til overflademontering af hærdningslys med dobbeltklæbende tapeOversigt over kontroller:

Foto

Trinvise instruktioner
Forberedelse

1) Oplad batterierne før hærdningslyset bruges (se afsnittet Batterivedligeholdelse).
2) Placer hærdningslyset i en standardkonsol til dentale enheder eller konsol til montering af tilbehør, indtil det er parat til brug.
3) Før hver brug skal du placere en ny barrieremuffe over hærdningslyset.
Den hygiejniske barrieremuffe er tilpasset hærdningslyset og holder hærdningslysets overflade ren. Barrieremuffen hjælper med at forhindre krydskontaminering, hjælper med atholde tandkompositmaterialet ved at klæbe til overfladen af linsen og hærdningslyset og forhindrer misfarvning og korrosion forårsaget af rengøringsopløsninger.

Bemærk:• Brug af den hygiejniske barrieremuffe reducerer lysudgangen med 5-10 %. På grund af hærdningslysets høje udgangseffekt, har hærdning vist sig at være i det væsentligeækvivalent.
• Hærdningslyset skal rengøres og desinficeres med passende rengørings- og/eller desinficeringsmidler efter hver patient. Se afsnittet Behandling.

Knapper og lys

Forklaring

Timerlys
Lavt batteri - Indikator
Tilstand/statuslys
Timer/Knappen Skift tilstand(Top)

For alle de beskrevne produkter skal du læse og forstå alle instruktioner og SDS-oplysninger omhyggeligt før brug.

Indikationer for brug/tilsigtet formål

Belysningskilde til hærdning af fotoaktiverede dentale retablerende materialer og klæbemidler.

Advarsler og forholdsregler
Risikogruppe 2
FORSIGTIG UV udsendes fra dette produkt. Øjen- eller hudirritation kan skyldes eksponering. Brug passende afskærmning.
FORSIGTIG Mulig farlig optisk stråling udsendes fra dette produkt. Stir ikke på lyset under brug. Kan være skadeligt for øjnene.
• Kig IKKE direkte ind i lysudgangen. Patient, kliniker og assistenter skal altid bære ravfarvet UV-øjenbeskyttelse, når hærdningslys er i brug.
• For at forhindre risikoen for elektrisk stød er det ikke tilladt at ændre dette udstyr. Brug kun den medfølgende Ultradent VALO strømforsyning og stikadaptere. Hvis disse
komponenter er beskadiget, må du ikke bruge dem, og du skal ringe til Ultradents kundeservice for at bestille en erstatning.
• Bærbart RF-kommunikationsudstyr kan nedbryde ydeevnen, hvis det bruges tættere på end 30 cm (12 in”).
• Brug kun autoriseret tilbehør, kabler og strømforsyninger for at forhindre ukorrekt brug, øgede elektromagnetiske emissioner eller nedsat elektromagnetisk immunitet (se afsnittet
Elektromagnetiske emissioner).
• For at undgå risikoen for elektrisk brand i forbindelse med håndtering af batterier:⸰ Bland IKKE genopladelige batterier med ikke-genopladelige batterier eller andre batterityper.
⸰ Forsøg IKKE at oplade ikke-genopladelige batterier.
⸰ Autoklaver IKKE og sprøjt ikke batterier, batterikontakter, opladere eller netadapter med væske af nogen art. Hvis der opstår korrosion på batteriopladerens kontakter, skal duringe til Ultradent Kundeservice for at bestille en udskiftning.
⸰ Oplad IKKE batterier omkring brandfarlige materialer.
⸰ Opbevar IKKE oplader i den kliniske operationsstue.
• For at undgå risiko for personskade, må du IKKE bruge batterier, der er korroderede (rust), har buler, udsender en lugt eller væske, har en revne eller manglende indpakning eller på
anden måde er beskadiget. Ring til Ultradent Kundeservice for at bestille udskiftningsbatterier.
• For at forhindre risikoen for termisk irritation eller skade, skal du undgå flere på hinanden følgende hærdningscyklusser, og oralt blødt væv må ikke eksponeres i umiddelbar nærhed
i mere end 10 sekunder i nogen tilstand. Hvis der kræves længere hærdningstider, skal du bruge flere korte hærdningscykler eller bruge et dobbelthærdende produkt for at undgå
opvarmning af blødt væv.
• Vær forsigtig ved behandling af patienter, der lider af uønskede fotobiologiske reaktioner eller følsomheder, patienter, der behandles med kemoterapi eller patienter, der behandles
med fotosensibiliserende medicin.
• Denne enhed kan være udsat for stærke magnetiske eller statiske elektriske felter, hvilket kan forstyrre programmeringen. Hvis du har mistanke om, at dette er sket, skal du straks
tage stikket ud af stikkontakten og derefter sætte den i stikkontakten igen.
• Tør IKKE hærdningslyset af med kaustiske eller slibende rengøringsmidler, autoklaver ikke og nedsænk ikke produktet i nogen form for ultralydsbad, desinfektionsmiddel,
rengøringsmiddel eller væske. Manglende overholdelse af medfølgende behandlingsinstruktioner kan gøre hærdningslyset ubrugeligt.
• For at undgå at beskadige udstyret, må du IKKE indsætte fingre, instrumenter eller andre genstande i batterirummet på hærdningslyset.
82Undgå at beskadige udstyret ved IKKE at rengøre guldkontakterne eller nogen del af batterirummet. Ring til Ultradent Kundeservice, hvis du har et spørgsmål.
For at forhindre risikoen for krydskontaminering, er barrieremuffer kun til brug på en enkelt patient.
For at reducere risikoen for korrosion skal du fjerne barrieremuffen efter brug.
For at reducere risikoen for underhærdede harpikser må du ikke bruge hærdningslys, hvis linsen er beskadiget.

Installation af hygiejniske barrieremuffer:Tænd/sluk-knap(per)(Toppen eller bunden)(Bund)••••VALO Trådløs Lysskærm:VALO Trådløs Lysskærm er ovalformet, kan roteres til maksimal brug, og kan bruges med en gennemsigtig barrieremuffe.

Brug1) Hver strømtilstand bruges til hærdning af dentale materialer med fotoinitiatorer. Se Lynvejledning om tilstande for anbefalede hærdningstider.
BEMÆRK: Hærdningslyset er programmeret til at cykle fra standardstrøm til High strøm Plus til Xtra strømtilstand i rækkefølge. Hvis du for eksempel skal skifte fra standardstrømtilstandtil Xtra strømtilstand, er det nødvendigt at cykle ind i High strøm Plus-tilstanden og derefter til Xtra strømtilstanden.
2) The curing light stores the most recentely used timing inverval and mode, and it will default back to this whenever the modes are changed or if the batteries are removed.

BetjeningHÆRDNINGSTILSTAND: Standardstrømtilstand

TIMERINTERVALLER: 5, 10, 15, 20 sekunder.
• Denne tilstand er standardindstilling for hærdningslyset, når det tændes FØRSTE GANG. Tilstands-/statuslyset bliver grønt, og de fire grønne timerlys tændes, hvilket angiverstandardstrømtilstand.
• For at ændre tidsintervaller, trykkes hurtigt på knappen Tid/tilstand.
• Tryk på tænd/sluk-knappen for at hærde. For at stoppe hærdningen inden afslutningen af et tidsinterval, tryk på tænd/sluk-knappen igen.

HÆRDNINGSTILSTAND: High strøm Plus-tilstand

TIMERINTERVALLER: 1, 2, 3, 4 sekunder.
• Fra standardstrømtilstand skal du trykke på og holde knappen Skift tid/tilstand i 2 sekunder og slip. Tilstands-/statuslyset vil være orange, og de fire grønne timerlys vil lyse og
blinke, hvilket angiver High strømtilstand.
• For at ændre tidsintervaller, trykkes hurtigt på knappen Tid/tilstand.
• Tryk på en af tænd/sluk-knapperne for at hærde. For at stoppe hærdningen inden færdiggørelsen af et tidsinterval, tryk på en af tænd/sluk-knapperne igen.
• For at vende tilbage til standardstrømtilstand, tryk og hold knappen Skift tid/tilstand i 2 sekunder og slip; dette vil cykle til Xtra strømtilstand. Tryk og hold igen i 2 sekunder, og slip.
Tilstands-/statuslyset bliver grønt, og de fire grønne timerlys lyser, hvilket angiver standardtilstand.

HÆRDNINGSTILSTAND: Xtra strømtilstand

TIMERINTERVAL: Kun 3 sekunder (Bemærk: Xtra strømtilstand har 2 sekunders sikkerhedsforsinkelse i slutningen af hver hærdningscyklus for at begrænse opvarmning under
fortløbende hærdning. Ved slutningen af forsinkelsen angiver bip, at apparatet er klar til fortsat brug).
• Fra standardstrømtilstand, tryk på knappen Skift tid/tilstand i 2 sekunder, slip, tryk og hold den nede i 2 sekunder, og slip. Tilstands-/statuslyset vil være orange og blinke, og tre af
de grønne timerlys vil lyse og blinke, hvilket angiver Xtra strømtilstand.
• Tryk på tænd/sluk-knappen for at hærde. For at stoppe hærdningen inden afslutningen af et tidsinterval, tryk på tænd/sluk-knappen igen.
• For at vende tilbage til standardstrømtilstand, skal du trykke og holde knappen Time/Mode (Tid/tilstand) i 2 sekunder og slip. Tilstands-/statuslyset bliver grønt, og de grønne
timerlys tændes, hvilket angiver standardstrømtilstand..

Dvaletilstand: Hærdningslyset går i dvaletilstand efter 60 sekunders inaktivitet, som angivet ved langsom blinkning af tilstands-/statuslyset. Hvis enheden samles op eller røres, vilhærdningslyset blive vågnet og automatisk returnere til den sidst anvendte indstilling. Lad hærdningslyset være uforstyrret, når det ikke er i brug, for at maksimere batteriets levetid.

Rengøring1) assér brugte barrieremuffer i standardaffald efter hver patient.
2) Se Behandlingssektion.

Instruktioner vedr. monteringskonsol

1) Konsollen skal monteres på en flad, oliefri overflade.
2) Rengør overfladen med hospitalssprit.
3) Træk bagsiden af konsollens klæbende tape.
4) Placer konsollen, så hærdningslyset løfter opad, når det fjernes. Tryk den let på plads.

83

Lynvejledning om tilstande:Standardstrøm
1000 mW/cm2

TilstandHigh strøm Plus
1600 mW/cm2

Xtra strøm
3200 mW/cm2

Tænd/sluk-knap
Tilstand/timer-LED'er

Tidsknapper

Tidsindstillinger

5 sek

10 sek15 sek20 sek

1 sek

2 sek

3 sek

4 sek

Sådan ændres tid

Tryk og slip tidsknappen hurtigt for at gå gennem tidsmuligheder.

Sådan ændres tilstande

Tryk og hold tasten Time (Tid) i 2 sekunder og slip. Hærdningslyset vil cykle til næste tilstand.

Forklaring

Faste LED'er

Kun 3 sek

Blinkende LED'er

Lave batterier: Hærdningslyset signalerer brugeren, at det er på tide at skifte batterierne, når indikatoren for lavt batteri blinker rødt. Hvis batteriladningen bliver for lav, udsendes deren hørbar advarselslyd med 3 bip, og hærdningslyset tillader ikke yderligere drift, før batterierne er opladet eller nye batterier er sat i. (Se Hurtig advarselsvejledning)

Curing Mode

Power* (mW)

Irradiance* (mW/cm2)

Total Exposure Time (Seconds)

Energy** (Joules)

Standard

970

900

5/10/15/20

4.8J/9.7J/14.5J/19.4J

High Power Plus

1,620

1,500

1/2/3/4

1.6J/3.2J/4.8J/6.5J

Xtra

2,260

2,100

3

6.8J

Udskift batterierne

Opladetid og batterilevetid: Fuldt opladet batterilevetid i hærdningslyset afhænger af tilstands-/tidsinterval, batteritype, brug og LED-effektivitet. Generelt skal genopladelige batteriervare 1 - 2 uger. Ikke-genopladelige batterier kan vare 2 - 3 gange længere.
• Anbefalet interval for genopladning: Når indikatoren for lavt batteri tænder, eller ca. hver 1 - 2 uger afhængigt af brug.
• Ekstra batterier: Hærdningslyset leveres med 4 genopladelige batterier. Vi anbefaler også, at der opbevares et ekstra sæt ikke-genopladelige CR123A batterier, hvis der opstår etproblem eller tab af de genopladelige batterier.
• Batterilevetid: Genopladelige lithium-jernfosfatbatterier kan genoplades ca. 1000 til 2000 gange. Baseret på normal brug og korrekt pleje bør batterierne vare op til fem år, men skal
udskiftes efter behov.

Garanti

Lynvejledning om advarsler:• Lavt batteri: Blinker langsomt• Slukning: 3 bip, blinker• Forbyder driftSådan skiftes/indsættes batterier:
1) Fjern bagdækslet ved at dreje mod uret en kvart omgang.
2) Fjern batterierne.
3) Indsæt nye batterier med den positive (+) side først.
4) Fastgør bagdækslet ved at justere og forsigtigt skubbe, mens du drejer med uret. Hætten vil klikke, når den er helt tilsluttet.
5) Enheden er klar til brug.
Ultradent godkender om nødvendigt følgende IKKE-GENOPLADELIGE batterier til hærdningslyset:BEMÆRK: Forsøg IKKE at oplade ikke-genopladelige batterier.
1) Tenergy Propel Photo Lithium2) Titanium Innovations CR123A
3) Energizer® 1234) Duracell® Ultra CR123A
5) SureFire® SF123A
6) Panasonic® CR123A

Lynvejledning om hærdning:

Advarsel om strømniveauSådan oplades batterierne:1) Sæt opladeren i stikkontakten.
2) Sæt batterierne i oplader med den positive (+) ende mod indikatorlysene på opladeren.
3) Grønne lys angiver, at batterierne er klar til brug.
4) Det tager 1-3 timer at oplade batterierne. Lad batterierne sidde i opladeren, indtil de er klar til brug.
BEMÆRK: Hvis det røde lys på opladeren ikke bliver grønt, når batterierne har opladet mere end tre timer, kan batteriet blive ødelagt (beskadiget) og kan ikke oplades. Prøv et nyt batteri
eller ring til Ultradent kundeservice for at bestille et nyt sæt genopladelige batterier.

Advarsel om temperatur
Lad dem afkøle• 3 bip• Blinker langsomt• Forbyder drift

Advarsel om service

LED-advarsel

For reparation, ring til kundeservice

For reparation, ring til kundeservice• Ingen lyd• Blinker, 2 sekunder• Tillader drift• 3 kontinuerlige bip• Enkelt hurtig blink• Forbyder drift

Ultradent garanterer hermed, at dette instrument overholder de specifikationer, som beskrives i Ultradents dokumentation, der ledsager produktet, i en periode på 5 år* i allevæsentlige henseender og være fri for enhver mangel i materialer eller håndværk. Denne garanti gælder udelukkende for den oprindelige køber og kan ikke overdrages. Alle defekte
produkter skal returneres til Ultradent. Der er ingen komponenter i VALO Trådløs hærdningslyssystem, der kan serviceres af bruger. Manipulation af VALO Trådløs hærdningslys vil
ugyldiggøre garantien.
Garantien for VALO Trådløs hærdningslys dækker ikke skader forårsaget af kunder. Hvis en VALO Trådløs for eksempel misbruges eller tabes og linsen går i stykker, er kunden ansvarlig
for at betale for eventuelle nødvendige reparationer.
*Med salgskvittering, der angiver datoen for salget til tandlægen.

Bearbejdning

Efter hver brug fugtes et stykke gaze eller en blød klud med et godkendt desinfektionsmiddel til overflader, og overfladen og linsen tørres af.
ACCEPTABLE RENGØRINGSMIDLER:• Lysol Brand III Desinfektionsspray (anbefalet)• Isopropylalkohol• Ethylalkoholbaserede rengøringsmidler• Lysol®* Koncentrat (kun alkoholbaseret)

Vedligeholdelse
Reparation

UReparation udført af bruger
1) Kontroller linsen rutinemæssigt for hærdet dental harpiks. Brug om nødvendigt et dentalt instrument uden diamant til omhyggeligt at fjerne eventuel fastklæbet harpiks.
2) Lysmålere er meget forskellige og er designet til specifikke lyslederspidser og linser. Ultradent anbefaler at kontrollere udgang rutinemæssigt i standardstrømtilstand. BEMÆRK: Densande numeriske udgang er skæv på grund af unøjagtigheden af almindelige lysmålere og den brugerdefinerede LED-pakke i hærdningslyset.
Reparation udført af producent
1) Reparationer må kun udføres af autoriseret servicepersonale. Ultradent skal give servicepersonale dokumentation for at de kan udføre reparationer.
2) Når du sender enheder til reparation, service eller kalibrering, skal du altid fjerne batterierne fra hærdningslyset og opladeren. Pak batterier, oplader, adapter og hærdelys separati returkassen.
3) Send batterier i overensstemmelse med lokale bestemmelser.

Vedligeholdelse af batteri

Opladning og ændring af batterierHærdningslyset leveres med 4 genopladelige lithium-jernfosfatbatterier.

84

UACCEPTABLE RENGØRINGSMIDLER - BRUG IKKE:• Stærkt alkalisk rengøringsmiddel af enhver art, herunder håndsæbe og opvaskemiddel• Blegemiddelbaserede rengøringsmidler (fx Clorox™*, Sterilox™*)• Hydrogenperoxidbaserede rengøringsmidler• Slibende rengøringsmidler (fx Comet Cleanser™*)• Acetone- eller kulbrintebaserede rengøringsmidler• MEK (methylethylketon)• Birex®*• Gluteraldehyd• Kvaternære ammoniumchlorid-saltbaserede rengøringsmidler• Cavicide1™*-opløsning eller servietters• Cavicide™* produkter (uden blegemiddel)***Varemærke, der tilhører et andet selskab end Ultradent** Hvis det bruges, kan det falme farven
BATTERIOPLADER:
Hvis rengøring bliver nødvendigt, skal du tage stikket ud af stikkontakten, fugte en klud med isopropylalkohol, og forsigtigt aftørre opladerens eller batteriernes overflade. Lad opladeren
tørre helt, inden produktet bruges igen.
85

Tekniske oplysninger/dataRENGØRING AF LYSSKÆRM::Kold-desinficer VALO Trådløs Lysskærm ved brug af ethvert disinfektionsmiddel til overflader. Autoklaver IKKE.

Bølgelængdebånd til effektiv hærdning af komposit:

Opbevaring og bortskaffelse

EFFEKTIVE KOMPOSITHÆRDENDE BÅLÆNGDEBÅND
(Anvendeligt bølgelængdeområde: 385-515 nm)

Hvis hærdningslyset opbevares i perioder længere end 2 uger eller det nedpakkes til kørsel, skal du altid fjerne batterierne. Hvis der efterlades batterier i enheden i længere tid uden
genopladning, kan de ophøre med at fungere eller blive umulige at oplade. Opbevar ikke batterier ved temperaturer over 60 °C (140°F) eller i direkte sollys.
SPEKTRAL STRÅLEKRAFT (mW/nm)

Opbevaring og transport af hærdningslys:• Temperatur: +10 °C til +40 °C (+50 °F til +104 °F)• Relativ luftfugtighed: 10 % til 95 %• Omgivende tryk: 500 hPa til 1060 hPa
Ved bortskaffelse af elektronisk affald (dvs. hærdningslys, opladere, batterier og strømforsyninger) skal du følge lokale retningslinjer for affald og genanvendelse.

Tekniske overvejelser
Tilbehør
Vare

VALO Barrieremuffer

VALO Trådløs Lysskærm

VALO Oplader

CE-oplysninger

MDSS GmbH
Schiffgraben 4130175 HanoverTysklandFremstillet af:
TIDI Products, LLC.
570 Enterprise Drive
Neenah, WI 54956
Fremstillet i USA

Distribueret af:
Ultradent Products Inc
505 West Ultradent Drive (10200
South)South Jordan, UT 84095
USA

Effektivt øvrebølgelængdebånd
440-480 nm

XtraHigh PowerStandard

WAVEL ENGTH (nm)EgenskabOplysning/specifikation

Linse

Diameter 11,7 mm

Bølgelængdeområde

• Brugbart bølgelængdeområde: 385 - 515 nm• Spidsbølgelængder: 395 - 415 nm og 440 - 480 nm
Sammenligningsoversigt for nominel strålingsemittering

VALO Batterier

Lysintensi-tetstabelMåleinstru-ment†* Demetron
L.E.D.
Radiometer† MARC
SpektrumanalysatorMålers åbning

7 mm

3.9 mmStandard-strøm(±10%)

1000 mW/cm²

High strøm Plus(±10%)

1600 mW/cm²

Xtra strøm (±10%)86

Effektiv lavere
bølgelængdebånd
395-415 nm‡ Gigahertz spektrumanalysator
Emission

Samlet effekt

15 mm

15 mm

900 mW/cm²

970 mW

1800 mW/cm2

1500 mW/cm²

1615 mW

3200 mW/cm²(+/-20%)2100 mW/cm²

2260 mW

Strålingsemittering vil variere baseretpå instrumentkapacitet, målemetode og
lysplacering.
† Demetron-radiometre og MARCspektrumanalysatorer bør kun bruges som
reference, da de har mindre åbninger end
VALO-hærdningslys.
* Demetron-radiometre bør kun bruges som
reference, da de har begrænsninger i effekt og
spektral respons.
‡ Strålingsemittering er i overensstemmelse
med ISO 10650, når der måles med en
Gigahertz-spektrumanalysator.

VALO Storhærdnings-lys

Klassificeringer: IEC 60601-1 (sikkerhed), IEC 60601-1-2 (EMC)Vægt:• Med batterier: 6 oz. (190 gram)• Uden batterier: 5 oz. (150 gram)
Dimension: (8 x 1,28 x 1,06) tommer, (203 x 32,5 x 27) mm

Strømfor-syning til
oplader

Udgang - 12 VDC ved 500 mA
Indgang - 100 VAC til 240 VAC
Ultradent P/N 5930 VALO Strømforsyning til oplader med internationale
stikindsatser

Klassificeringer: IEC 60601-1 (sikkerhed)Ledningslængde - 1,8 meter (6 fod)
VALO Strømforsyning til oplader er en klasse II strømforsyning af
medicinsk kvalitet og yder isolation fra netstrømmen

87

Egenskab

Oplysning/specifikation

Vejlednings- og fremstillingserklæring for elektromagnetiske emissioner
Emissionstest

Overensstemmelse

Elektromagnetisk miljø - vejledning

RF-emissioner
CISPR 11

Gruppe 1Hærdningslyset anvender batterier og påvirkes ikke af EMI, RF eller overspændingsbeskyttelse.

VALO Oplader

VALO smart oplader til 3,6 VDC lithium-jernfosfatbatterier:• Automatisk slukning, når fuldt opladet• Automatisk registrering af defekte batterier• Beskyttelse: Termisk, overbelastning, kortslutning, omvendt polaritet
⸰ Rød LED - Oplader⸰ Grøn LED - Tom eller fuldt opladet⸰ LED slukket - kortslutning• Opladningstid: 1 – 3 timer
Klassificering: CE, WEEE

RF-emission
CISPR 11

Klasse BHærdningslyset bruger kun elektrisk og elektromagnetisk energi til deres interne funktioner.
Derfor er eventuelle RF-emissioner meget lave og kan ikke forårsage interferens på elektronisk
udstyr i nærheden.

N/A

Genopladelig: Sikker kemi lithium-jernfosfat (LiFePO4) RCR123A• Arbejdsspænding: 3,2 VDCRKlassificeringer: UL, CE, RoHS, WEEE

Harmoniske emissioner
IEC 61000-3-2

Harmonisk emission og spændingsvariation testning gælder ikke for hærdningslyset, fordi det
er batteridrevet.

VALO BatterierSpændingsfluktuationer/
flimmerudslip IEC 61000-3-

N/A

Driftsbetin-gelser

Temperatur: +10 °C til +32 °C (+50 °F til +90 °F)
Relativ luftfugtighed: 10 % til 95 %Omgivende tryk: 700 hPa til 1060 hPaHærdningslyset er egnet til brug i alle omgivelser, herunder private hjem og steder, som er
direkte forbundet med det offentlige lavspændingsnetværk, der leverer strøm til bygninger til
husholdningsbrug.

DriftscyklusHærdningslyset er beregnet til kortvarig drift. Ved maksimal omgivelsestemperatur (32 °C) 1 minut ON flere på hinanden følgende cykler, 30
minutter OFF (afkølingstid).

Fejlfinding
Hvis løsningerne foreslået nedenfor ikke afhjælper problemet, skal du ringe til Ultradent på 800.552.5512. Hvis du er uden for USA, skal du ringe til din Ultradentdistributør eller forhandler af dentale produkter.
ProblemMulige løsninger

Vejlednings- og fremstillingserklæring for elektromagnetisk immunitetHærdningslyset er beregnet til brug i det elektromagnetiske miljø, der er angivet nedenfor. Kunden eller brugeren skal sikre sig, at den bruges i et sådant miljø.
IMMUNITETS-TEST

IEC 60601 testniveau

Overensstemmelses-niveau

Elektromagnetisk miljø - vejledning

Elektrostatisk afladning (ESD)± 8 kV kontakt± 15 kV luft

± 8 kV kontakt± 15 kV luft

Fysisk miljø bør begrænses til følgende:
1) IP-kode: IP20
2) Nedsænk ikke i væske.
3) Må ikke anvendes omkring brandfarlig gas.
Enheden er ikke-APG og ikke-AP.
4) Fugtighedsinterval ved opbevaring: 10 % - 95 %5) Temperaturinterval ved opbevaring: 10 °C - 40 °C

IEC 61000-4-2

Lyset tændes ikke

1) Tryk på Time/Mode (Tid/tilstand) eller tænd/sluk-knappen for at vågne fra strømsparetilstand.
2) Kontroller den røde indikator for lavt batteriniveau for batteriladningsstatus.
3) Kontroller, at friske batterier er sat korrekt ind i enheden.
4) Hvis røde og gule LED-advarsler blinker, betyder det, at hærdningslyset har nået den indre temperatursikkerhedsgrænse.
Lad hærdningslyset køle ned i 10 minutter eller brug et køligt fugtigt håndklæde for at afkøle enheden hurtigt.
5) Hvis den røde LED-advarsel blinker og bipper kontinuerligt, skal du ringe til Ultradent Kundeservice til reparation.

Elektriske hurtige spændingsvarianter/strømstød
IEC 61000-4-4

Lyset forbliver ikke tændt i det ønskede
tidsrum

1) Kontroller tilstand og timerlys for korrekt tidsindgang.
2) Kontroller indikatoren for lavt batteriniveau for batteriladningsstatus.
3) Kontrollér, at de nye batterier er sat korrekt ind i enheden.

± 2 kV for strømforsyningsled± 2 kV for strømforsyningsninger
ledninger± 1 kV for indgangs-/udgangslinjer Bemærkning 1: Hærdningslyset
har ingen I/0 porte

Overspænding
IEC 61000-4-5± 1 kV linje til linje± 2 kV linje til jord

Lyset hærder ikke harpiks korrekt

1) Tjek linsen for resterende hærdet harpiks/komposit.
2) Brug korrekt gulbrun UV-øjenbeskyttelse og kontroller, at LED-lysene virker.
3) Kontroller effektniveauet med lysmåleren. Ved brug af en lysmåler anbefaler Ultradent at kontrollere hærdningslyset istandardstrømtilstand.
BEMÆRK: Den sande numeriske udgang vil blive skæv på grund af unøjagtigheden af almindelige lysmålere og den
brugerdefinerede LED-pakke, som hærdningslyset anvender. Lysmålere er meget forskellige og er designet til specifikke
lyslederspidser og linser.
4) Kontroller udløbsdatoen på hærdende harpiks.
5) Sørg for, at korrekt teknik følges efter producentens anbefalinger.

Batterier oplades ikke

1) Sørg for, at batterierne sættes i opladeren i den korrekte retning og lad batterierne oplade i 1-3 timer.
2) Hvis de røde lys på opladeren ikke skifter til grønt, skal du ringe til Ultradent Kundeservice for at bestilleerstatningsbatterier og/eller -opladere.
3) Hvis der ikke er synlige grønne eller røde lys på opladeren, skal du ringe Ultradent kundeservice for at bestille eller
udskifte oplader og/eller vekselstrømsadapter.

Opladeren oplader ikke batterierne

1) Sørg for, at opladeren er tilsluttet, og at vekselstrømsadapteren er tilsluttet en stikkontakt.
2) Hvis de grønne eller røde lys på opladeren ikke er synlige, skal du ringe til Ultradent Kundeservice for at få en ny oplader
og/eller vekselstrømsadapter.

Kan ikke ændre tilstand eller
tidsintervaller

1) Hold både knapperne Time/Mode (Tid/tilstand) og Tænd/sluk-knap nede, indtil en række bip angiver, at hærdningslyseter låst op.

Hærdningslyset er batteridrevet og kan ikke tilsluttes
AC MAINS-strøm.

± 1 kV linje til linje± 2 kV linje til jord

Fordi hærdningslyset er batteridrevet, er det ikke
underlagt elektriske transienter, overspændinger,spændingsfald, kortslutninger, afbrydelser eller
variationer på AC MAINS-strømmen.

Spænding, fald, kortslutninger,<5%U
afbrydelser og variationer på(> 95 % fald i U ved 0,5 cyklus)
strømforsynin-gens indgangslinjer
40 % UIEC 61000-4-11(60 % fald i U ved 5 cyklusser)<5%U(> 95 % fald i U ved 0,5 cyklus)

Den ekstra batterioplader kan være underlagt
ovenstående, men den er adskilt fra og ikke kritisk fordriften af VALO Trådløs hærdningslys.

70 % U(30 % fald i U ved 25 cyklusser)

70 % U(30 % fald i U ved 25 cyklusser)<5%U(> 95 % fald i U ved 5 sek)

<5%U(> 95 % fald i U ved 5 sek)Bemærkning 2: Genopretter selv

30 A/m30 A/mStrømfrekvens(50/60 Hz) magnetfelt

40 % U(60 % fald i U ved 5 cyklusser)

IEC 61000-4-8

Hvis batterispændingen i hærdningslyset falder til
4 VDC, tillader apparatet ikke drift. VALO Trådløs
hærdningslys slukker. Når nye batterier sættes i og
de korrekte effektniveauer genoprettes, genoptages
hærdningslyset og vender tilbage til samme tilstand,før strømforbruget. Hærdningslyset vil selvstarte, hvis
strømmen tabes.
Magnetfelter med magnetfrekvens bør være på
niveauer, der karakteriseres af en typisk placering iet typisk bolig-, hjemmepleje-, kommercielt, hospitaleller militært miljø.

Bemærkning 1: Hærdningslyset er ikke udstyret med porte eller tilgængelige I/O-linjer.

Diverse oplysninger
Vejlednings- og fremstillingserklæring for elektromagnetiske emissionerHærdningslyset er beregnet til brug i det elektromagnetiske miljø, der er angivet nedenfor. Kunden eller brugeren skal sikre sig, at den bruges i et sådant miljø.
88

89

Vejlednings- og fremstillingserklæring for elektromagnetisk immunitet for systemer, der ikke er livsopretholdende

Vejlednings- og fremstillingserklæring for anbefalede afstande mellem bærbart og mobilt RF-kommunikationsudstyr og VALO Trådløs hærdningslysHærdningslyset er beregnet til brug i det elektromagnetiske miljø, der er angivet nedenfor. Kunden eller brugeren skal sikre sig, at den bruges i et sådant miljø.

Hærdningslyset er beregnet til brug i et elektromagnetisk miljø, hvor udstrålede RF-forstyrrelser styres. Brugeren af hærdningslyset kan medvirke til at forhindreelektromagnetisk interferens ved at opretholde en minimal afstand mellem bærbart og mobilt RF-kommunikationsudstyr (sendere) og hærdningslyset, som
anbefalet nedenfor i henhold til kommunikationsudstyrets maksimale udgangseffekt.

IMMUNITETS-TEST

IEC 60601 testniveau

Overensstemmelsesniveau

Konduktion RF

3 Vrms3 VrmsIEC 61000-4-6

150 kHz til 80 MHz

150 kHz til 80 MHz

Elektromagnetisk miljø - vejledningBærbart og mobilt RF-kommunikationsudstyr skal ikke anvendes tættere på nogen del afhærdningslyset, herunder kabler, end den anbefalede adskillelsesafstand, beregnet ud fra
den ligning, der gælder for senderens frekvens.
Anbefalet adskillelsesafstand::80 MHz til 800 MHz

800 MHz til 2,5 GHz
Udstrålet RF

3 V/m

3 V/m

P er senderens maksimale udgangseffekt i watt (W) ifølge senderens producent, og d er
den anbefalede separationsafstand i meter (m).
Feltstyrker fra faste RF-sendere, som bestemt af en elektromagnetisk undersøgelsea, børvære mindre end overensstemmelsesniveauet i hvert frekvensområdeb.

IEC 61000-4-3

80 MHz til 2,5 GHz

80 MHz til 2,5 GHzInterferens kan forekomme i nærheden af udstyr mærket med følgende symbol:

Separationsafstand afhængig af senderens frekvens(meter)Nominel maksimal udgangseffekt
af senderen(P i watt)

150 kHz – 80 MHz

80 MHz – 800 MHz

800 MHz – 2,5 GHz

0,01

0,12 meter

0,035 meter

0,07 meter

0,1

0,37 meter

0,11 meter

0,22 meter

1

1,7 meter

0,35 meter

0,7 meter

10

3,7 meter

1,11 meter

2,22 meter

100

11,7 meter

3,5 meter

7,0 meterHærdningslyset er blevet testet i henhold til IEC 60601-1-2: 2014 og bestået under udstrålede feltstyrker på 10 V/m mellem 80 MHz til 2,5 GHz. Værdien af 3 Vrms
svarer til V1, og værdien 10 V/m svarer til E1 i formlerne ovenfor.
For sendere, der er bedømt til en maksimal udgangseffekt, som ikke er anført ovenfor, kan den anbefalede adskillelsesafstand d i meter (m) estimeres ved hjælp af
den ligning, der gælder for frekvensen af senderen, hvor P er senderens maksimale udgangseffekt i watt (W) ifølge producenten af senderen.
BEMÆRKNING 1 Ved 80 MHz og 800 MHz gælder separationsafstanden for det højere frekvensområde.
BEMÆRKNING 2 Disse retningslinjer finder måske ikke anvendelse i alle situationer. Elektromagnetisk forplantning påvirkes af absorption og refleksion fra strukturer,
objekter og mennesker.

BEMÆRKNING 1 Ved 80 MHz og 800 MHz gælder det højere frekvensområde.
BEMÆRKNING 2 Disse retningslinjer finder måske ikke anvendelse i alle situationer. Elektromagnetisk forplantning påvirkes af absorption og refleksion fra strukturer,
objekter og mennesker.
aFeltstyrker fra faste sendere, som fx basestationer til radio (mobiltelefoner/trådløse) telefoner og landmobile radioer, amatørradio, AM- og FM-radiotransmission
og tv-transmission, kan ikke forudsiges teoretisk med nøjagtighed. For at vurdere det elektromagnetiske miljø, der skyldes faste RF-sendere skal en elektromagnetisk
undersøgelse af stedet overvejes. Hvis den målte feltstyrke på det sted, hvor hærdningslyset bruges, overstiger det gældende RF-overensstemmelsesniveau ovenfor,skal VALO Trådløs hærdningslys observeres, så normal drift kan bekræftes. Hvis unormal ydeevne observeres, kan der være behov for yderligere foranstaltninger,
som f.eks. reorientering eller omplacering af hærdningslyset.
bOver frekvensområdet 150 kHz til 80 MHz bør feltstyrker være mindre end 3 V/m.

90

91

VALO™ GRANDFIVALOKOVETIN

Tuotekuvaus

Laajan aallonpituus spektrin ansiosta, VALO Grand -laite kykenee polymeroimaan kaikki valokovetteiset tuotteet, joiden aallonpituusalue on 385-515 nm:n välillä ISO 10650 -standardin
mukaisesti. VALO Grand -valokovetin käyttää uudelleen ladattavaa Ultradent VALO -akkua ja -laturia. Valokovetin on suunniteltu lepäämään hammashoitoyksikön standardikannattimessa
tai se voidaan asentaa mukana toimitettuun räätälöityyn pidikkeeseen.

VALO Grand -laitteen osat:
•
•
•
•
•
•

Hygieenisen suojan asentaminen:• Hygieenisen suojan käyttäminen vähentää valotehoa 5-10%. Valokovettimen korkean lähtötehon ansiosta, kovetuksen on todettu pysyvän merkittävän samanarvoisena.
• Valokovetin on puhdistettava ja desinfioitava asianmukaisilla pesu- ja/tai puhdistusaineilla kunkin potilaan jälkeen. Katso osio Käsittely.

Johdottoman VALO -laitteen valosuoja:

Johdottoman VALO -laitteen valosuoja on soikea, sitä voi käännellä hyödyn maksimoimiseksi ja käyttää läpinäkyvän suojapussin kanssa.

Painikkeet ja valot

SeliteVirtakytkimet(Ylä- tai alaosa)Ajoitus valot
Alhainen akun varaustaso -ilmaisin
TilavaloAjoituksen/tilan
vaihtopainike

Käyttö

1) Kutakin tehotilaa käytetään hammasmateriaalien kovettamiseen valokäynnisteen avulla. Katso Tilojen Pikaopas suositeltuihin kovettumisaikoihin.
HUOMAUTUS: Valokovetin on ohjelmoitu siirtymään järjestyksessä Normaali teho -tilasta Korkea Plus teho -tilaan, ja sen jälkeen Erittäin korkea teho -tilaan. Esimerkiksi, vaihtaaksesi
Normaali teho -tilasta Erittäin korkea teho -tilaan, joudut siirtymään sinne Korkea Plus teho -tilan kautta.
2) Valokovetin tallentaa viimeksi käytetyn ajoituksen aikavälin ja tilan, ja palaa takaisin niihin, jos tilaa vaihdetaan tai akut poistetaan.

Käyttö
KOVETUSTILA: Normaalin tehon tila

AJOITUKSEN AIKAVÄLIT: 5, 10, 15, 20 sekuntia.
• Valokovettimen oletusasetuksena on tämä tila, kun se kytketään päälle ensimmäisen kerran. Tilan merkkivalo palaa vihreänä ja neljä vihreää ajoituksen valoa palavat, ilmaisten
Normaali teho -tilan.
• Muuttaaksesi ajoituksen väliä, paina nopeasti Aika/Tila -painiketta.
• Paina virtapainiketta kovettaaksesi. Kovetuksen lopettamiseen ennen ajoitusvälin loppumista, paina virtapainiketta uudelleen.

KOVETUSTILA: Korkea Plus teho -tila(Ylä)

Lue kaikki ohjeet sekä käyttöturvallisuustiedote huolellisesti ja sisäistä ne ennen kyseisten tuotteiden käyttöä.

Käyttöaiheet/-tarkoitus

Valoaktivoituvien hampaiden korjausmateriaalien ja liimojen kovetukseen käytettävä valolähde.

Varoitukset ja varotoimet
Riskiryhmä 2
VAROITUS - Tämä tuote lähettää UV-säteitä. Altistuminen voi aiheuttaa silmien tai ihon ärsytystä. Käytä asianmukaista suojausta.
VAROITUS - Tämä tuote lähettää mahdollisesti vaarallista optista säteilyä. Älä tuijota käyttövaloa. Saattaa olla haitallista silmille.
• ÄLÄ katso suoraan valonlähteeseen. Valokovetinta käytettäessä, potilaan, lääkärin ja avustajien tulisi aina käyttää oransseja UV-suojalaseja.
• Sähköiskun vaaran välttämiseksi tähän laitteeseen ei saa tehdä mitään muutoksia. Käytä vain mukana toimitettua Ultradent VALO -virtalähdettä ja -pistokeliitintä. Jos nämä
komponentit ovat vaurioituneet, älä käytä ja soita Ultradent-asiakaspalveluun tilataksesi korvaavan osan.
• Kannettavat RF-viestintälaitteet voivat heikentää suorituskykyä, jos niitä käytetään lähempänä kuin 30 cm (12 tuumaa).
• Käytä vain hyväksyttyjä lisävarusteita, kaapeleita ja virtalähteitä väärinkäytön, lisääntyneiden sähkömagneettisten päästöjen tai sähkömagneettisen häiriön vähentämiseksi (katsokohta Sähkömagneettiset päästöt).
• Välttääksesi akkujen käsittelyyn liittyvän sähköpalon:⸰ ÄLÄ käytä ladattavia akkuja kertakäyttöisten tai toisentyyppisten akkujen kanssa.
⸰ ÄLÄ yritä ladata kertakäyttöisiä paristoja.
⸰ ÄLÄ laita akkuja autoklaaviin, äläkä suihkuta niitä, niiden kosketinosia, laturia tai sen muuntajaa minkäänlaisella nesteellä. Jos akkulaturin koskettimissa ilmenee korroosiota, otayhteyttä Ultradentin asiakaspalveluun tilataksesi korvaavan tuotteen.
⸰ ÄLÄ lataa akkuja syttyvien materiaalien läheisyydessä.
⸰ ÄLÄ pidä laturissa kliinisen käytön aikana.
• Loukkaantumisriskin välttämiseksi, ÄLÄ käytä akkuja, jotka ovat syöpyneet (ruoste), lommoutuneet, erittävät hajua tai nesteitä, joiden suojakääre on repeytynyt tai puuttuu, tai ovat
muuten vahingoittuneita. Soita Ultradentin asiakaspalveluun ja tilaa vaihtoakku.
• Välttääksesi termisen ärsytyksen tai loukkaantumisen riskin, vältä peräkkäisiä kovetusjaksoja, äläkä altista suun pehmytkudoksia valon läheisyyteen yli 10 sekunnin ajan missään
toimintatilassa. Jos pidempi kovettumisaika on tarpeen, käytä useita lyhyempiä kovetusjaksoja tai käytä kaksoiskovetustuotetta pehmytkudoksen kuumentumisen välttämiseksi.
• Ole varovainen, kun hoidat potilaita, jotka kärsivät haitallisista fotobiologisista reaktioista tai herkkyyksistä, ovat kemoterapia hoidossa, tai käyttävät valolle herkistävää lääkitystä.
• Tämä yksikkö voi olla herkkä voimakkaille magneettisille tai staattisille sähkökentille, jotka voivat häiritä ohjelmointia. Jos epäilet, että näin on tapahtunut, irroita virtalähde hetkellisesti
ja kytke se sitten takaisin pistorasiaan.
• ÄLÄ pyyhi valokovetinta emäksisillä tai hankaavilla puhdistusaineilla, äläkä laita sitä autoklaaviin tai upota mihinkään ultraäänikylpyyn, desinfiointiaineeseen, puhdistusliuokseen tai
nesteeseen. Mukana olevien käsittelyohjeiden noudattamatta jättäminen voi tehdä valokovettimen käyttökelvottomaksi.
• Välttääksesi laitteen vahingoittumisen, ÄLÄ työnnä sormia, instrumentteja tai muita esineitä valokovettimen paristokoteloon.
• Välttääksesi laitteen vahingoittumisen, ÄLÄ yritä puhdistaa kultaisia kosketuspintoja tai paristokotelon muita osia. Jos on aihetta huoleen, soita Ultradentin asiakaspalveluun.
• Ristikontaminaatioriskin estämiseksi suojapussit on tarkoitettu kertakäyttöön.
921) Lataa akut ennen valokovettimen käyttöä (katso kohta Akun huolto).
2) Aseta valokovetin hammashoitoyksikön standardikannattimeen tai lisävarusteena toimitettuun pidikkeeseen, kunnes sitä tarvitaan käyttöön.
3) Ennen jokaista käyttökertaa, aseta uusi suojaholkki valokovettimen päälle.

Huomautus:

Kuva

(Ala)

Vaiheittaiset ohjeet
Valmistelutoimenpiteet

Hygieeninen suojapussi on räätälöity valokovettimelle ja pitää sen pinnan puhtaana. Suoja auttaa estämään ristikontaminaatiota, hampaiden komposiittimateriaalin tarttumista linssin
pintaan tai valokovettimen varteen, sekä puhdistusaineiden aiheuttamaa värjäytymistä ja korroosiota.

1 – VALO Grand -valokovetin
4 – Ultradent VALO uudelleen ladattavaa akkua
1 – Ultradent VALO -akkulaturi ja lääketieteellistä tasoa oleva 12 V:n verkkovirtamuuntaja
1 – VALO holkki -näytepakkaus
1 – VALO Grand -laitteen valosuoja
1 – Valokovettimen pinta-asennettava pidike ja kaksipuolinen teippi

Hallintalaitteiden yleiskatsaus:• Korroosioriskin vähentämiseksi poista suojapussi käytön jälkeen.
• Älä käytä kovetusvaloa, jos linssi on vaurioitunut, jotta vältetään alikovettuneiden materiaalien riski.

AJOITUKSEN AIKAVÄLIT: 1, 2, 3, 4 sekuntia.
• Normaali teho -tilassa, paina Aika/Tila -painiketta 2 sekunnin ajan ja vapauta. Tilan merkkivalo palaa oranssina ja neljä vihreää ajoituksen valoa syttyvät ja vilkkuvat, ilmaisten Korkea
teho -tilan.
• Muuttaaksesi ajoituksen väliä, paina nopeasti Aika/Tila -painiketta.
• Paina jompaakumpaa virtapainiketta kovettaaksesi. Kovetuksen lopettamiseen ennen ajoitusvälin loppumista, paina jompaakumpaa virtapainiketta uudelleen.
• Palataksesi Normaali teho -tilaan, paina Aika/Tila -painiketta 2 sekunnin ajan ja vapauta, laite siirtyy täten Erittäin korkea teho -tilaan. Paina Aikapainiketta 2 sekunnin ajan ja vapauta.
Tilan merkkivalo palaa vihreänä ja neljä vihreää ajoituksen valoa palavat, ilmaisten Normaali teho -tilan.

KOVETUSTILA: Erittäin korkean tehon tila

AJOITUKSEN AIKAVÄLI: Vain 3 sekuntia (Huomaa: Eritääin korkea teho -tilassa on 2 sekunnin turvaviive jokaisen kovetusjakson lopussa, joka rajoittaa kuumenemista peräkkäisten
kovetusten aikana. Viiveen lopussa, äänimerkki ilmoittaa että laite on valmis käytön jatkamiseen).
• Normaali teho -tilassa, paina Aika/Tila -painiketta 2 sekunnin ajan ja vapauta, paina toiset 2 sekuntia ja vapauta. Tilan merkkivalo palaa oranssina ja villkkuu ja kolme vihreää
ajoituksen valoa syttyy ja vilkkuu, ilmaisten Erittäin korkea teho -tilan.
• Paina virtapainiketta kovettaaksesi. Kovetuksen lopettamiseen ennen ajoitusvälin loppumista, paina virtapainiketta uudelleen.
• Palataksesi Normaali teho -tilaan, paina Aika/Tila -painiketta 2 sekunnin ajan ja vapauta. Tilan merkkivalo palaa vihreänä ja vihreät ajoituksen valot palavat, ilmaisten Normaali teho-tilan.

Lepotila: Valokovetin siirtyy lepotilaan 60 sekunnin toimettomuuden jälkeen ja se osoitetaan tila valon hitaalla vilkkumisella. Laitteen kosketus tai nostaminen herättää laitteen ja
palauttaa sen automaattisesti viimeksi käytettyyn asetukseen. Pidentääksesi akun kestoa, anna laitteen olla rauhassa kun sitä ei käytetä.

Puhdistus

1) Heitä käytetyt suojat sekajätteisiin kunkin potilaan jälkeen.
2) Katso Käsittely-osio.

Pidikkeen kiinnitysohjeet

1) Pidike tulee asentaa tasaiselle, öljy-vapaalle pinnalle.
2) Puhdista pinta spriillä.
3) Irroita pidikkeen teipin taustapaperi.
4) Aseta pidike siten, että valokovetinta nostetaan siirrettäessä ylöspäin. Paina tiukasti paikoilleen.

93

Tilojen Pikaopas:
Normaali teho
1000 mW/cm2

Tila

Korkea Plus teho
1600 mW/cm2

Erittäin korkea teho
3200 mW/cm2

Virtapainike
Tila/ Ajastus LEDit

Aika-painikkeet

Aika-asetukset

5s

10s

15s

20s

1s

2s

3s

Ajan muuttaminen

Paina ja vapauta Aikapainike nopeasti selataksesi aika vaihtoehtoja.

Tilan vaihtaminen

Paina Aikapainiketta 2 sekunnin ajan ja vapauta. Valokovetin siirtyy seuraavaan tilaan.

SeliteKiinteät LEDit

4s

Vain 3s

Vilkkuvat LEDit

Curing Mode

Power* (mW)

Irradiance* (mW/cm2)

Total Exposure Time (Seconds)

Energy** (Joules)

Standard

970

900

5/10/15/20

4.8J/9.7J/14.5J/19.4J

High Power Plus

1,620

1,500

1/2/3/4

1.6J/3.2J/4.8J/6.5J

Xtra

2,260

2,100

3

6.8J

Vaihda akut

1) Tenergy Propel Photo Lithium2) Titanium Innovations CR123A
3) Energizer® 1234) Duracell® Ultra CR123A
5) SureFire® SF123A
6) Panasonic® CR123A

Latausaika ja akun kesto: Täyteen ladatun akun kesto valokovettimessa riippuu sen Tila-asetuksesta, ajastuksen aikavälistä, akkutyypistä, käytön määrästä, ja LEDien tehokkuudesta.
Yleensä ladattavat akut kestävät 1-2 viikkoa. Kertakäyttäiset paristot voivat kestää 2-3 kertaa pidempään.
• Suositeltava aikaväli latauksille: Kun akun merkkivalo syttyy tai suunnilleen 1-2 viikon välein, riippuen käytöstä.
• Vara-akut: Valokovetin toimitetaan 4 uudelleen ladattavan akun kanssa. Suosittelemme pitämään käsillä myös muutamia ei-ladattavia CR123A-paristoja ongelmatilanteiden tai
ladattavien akkujen menetyksen varalta.
• Akun käyttöikäarvio: Ladattavat litium-rautafosfaattiakut voidaan ladata noin 1000–2000 kertaa. Normaalikäytössä ja asianmukaisesti hoidettuna akut saattavat kestää jopa viisi
vuotta, mutta tulisi vaihtaa tarpeen vaatiessa.

Takuu

Varoitusten Pikaopas:• Akun varaustaso alhainen: hidas
vilkkuminen• Sammuta: 3 äänimerkkiä, vilkkuva• Estää toiminnan

Akkujen vaihto/paikalleen asettaminen:
1) Poista kansi kiertämällä sitä vastapäivään neljäsosa kierroksen verran.
2) Poista akut.
3) Aseta uudet akut positiivinen (+) napa edellä.
4) Kiinnittääksesi kannen paikoilleen, kohdista ja työnnä sitä samalla kun kierrät myötäpäivään. Kansi klikkaa, kun se on täysin paikoillaan.
5) Laite on käyttövalmis.
Tarvittaessa, Ultradent valtuuttaa seuraavien EI-LADATTAVIEN paristojen käytön valokovettimessa:
HUOMAUTUS: ÄLÄ yritä ladata kertakäyttöisiä paristoja.

Alhainen akun varaustaso: Valokovetin viestittää käyttäjälle, että on aika vaihtaa akut, kun Alhainen akun varaustaso -merkkivalo vilkkuu punaisena. Jos akun varaus laskee liian alhaiseksi,
kuuluu 3:n piippauksen varoitusääni, joka ilmaisee että valokovetin ei salli muita toimintoja ennen kuin akut ovat latautuneet tai laitteeseen on asennettu uudet akut. (Katso Varoitusten
Pikaopas)

Kovetuksen Pikaopas:

Tehotaso varoitus

Akkujen lataaminen:
1) Kytke laturi pistorasiaan.
2) Aseta akut laturiin niin, että positiivinen (+) napa osoittaa laturin merkkivaloja kohti.
3) Vihreät valot osittavat, että akut ovat käyttövalmiita.
4) Akkujen lataaminen kestää 1-3 tuntia. Jätä akut laturiin, kunnes niitä tarvitaan käytössä.
HUOMAUTUS: Jos punainen latausvalo ei muutu vihreäksi akun ladattua yli kolme tuntia, akku saattaa olla heikentynyt (vioittunut) eikä ole enää ladattavissa. Kokeile toista akkua tai soita
Ultradentin asiakaspalveluun ja tilaa uusi erä ladattavia akkuja.

Lämpötila varoitus
Anna jäähtyä
• 3 äänimerkkiä• Hidas vilkkuminen• Estää toiminnan

Huoltovaroitus

LED varoitus

Soita asiakaspalveluun korjausta varten Soita asiakaspalveluun korjausta varten• Ei ääntä
• Vilkkuu, 2 sekuntia• Sallii toiminnan• Jatkuva 3 piippausta• Yksi nopea vilkahdus• Estää toiminnan

Ultradent takaa 5 vuoden ajan*, että laite täyttää kaikilta olennaisilta osiltaan Ultradentin mukana tulevissa asiakirjoissa esitetyt spesifikaatit, eikä siinä ole materiaalivikoja tai
valmistusvirheitä. Tämä takuu koskee ainoastaan alkuperäistä ostajaa, eikä sitä voi siirtää. Kaikki vialliset tuotteet tulee palauttaa Ultradentille. Johdottomassa VALO -järjestelmässä ei ole
asiakkaan huolettavissa olevia osia. Johdottoman VALO -laitteen peukalointi mitätöi takuun.
Johdottoman VALO -laitteen takuu ei kata asiakkaiden aiheuttamia vahinkoja. Esimerkiksi; jos Johdotonta VALO -laitetta käytetään väärin tai se putoaa ja linssi rikkoutuu, asiakas on
velvollinen korvaamaan kaikki tarpeelliset korjaukset.
*Vaatii ostokuitin, joka osoittaa myyntipäivän hammaslääkärille.

Käsittely

Jokaisen käyttökerran jälkeen, kostuta sideharso tai pehmeä kangas hyväksytyllä pintadesinfektioiaineella ja pyyhi pinnat ja linssit.
HYVÄKSYTYT PUHDISTUSAINEET:• Lysol Brand III -desinfiointisuihke (suositeltava)• Isopropyylialkoholi• Etyylisprii puhdistusaineet• Lysol®*-tiiviste (vain alkoholipohjainen)

HuoltoKorjaus

Käyttäjän suorittamat korjaukset
1) Tarkasta linssit säännöllisesti kovettuneen sidosaineen varalta. Tarvittaessa, poista tarttunut sidosaine varovaisesti timantteja sisältämättömillä instrumenteilla.
2) Valomittarit poikkeavat toisistaan merkittävästi, ja ne on suunniteltu tietyille valonohjauskärjille ja linsseille. Ultradent suosittelee laitteen tehon säännöllistä tarkastamista Normaaliteho -tilassa. HUOMAUTUS: todellinen numeerinen teho vääristyy yleisten valomittareiden epätarkkuuden ja valokovettimen käyttämän erikoisvalmisteisen LED-pakkauksen johdosta.
Valmistajan suorittamat korjaukset
1) Korjaukset saa suorittaa vain valtuutettu huoltohenkilöstö. Ultradent toimittaa huoltohenkilökunnalle korjaukseen tarvittavat asiakirjat.
2) Aina kun lähetät laitetta korjattavaksi, huoltoon tai kalibrointia varten, poista akut valokovettimesta ja laturista. Pakkaa akut, laturi, adapteri ja valokovetin erikseen palautuslaatikkoon.
3) Lähetä akut paikallisten säännösten mukaisesti.

Akun huolto

Akkujen lataaminen ja vaihtaminen
Valokovetin toimitetaan 4 ladattavan litium-rautafosfaattiakun kanssa.

94

SOPIMATTOMAT PUHDISTUSAINEET - ÄLÄ KÄYTÄ:• Vahvasti emäksisiä pesuaineita, mukaan lukien käsisaippuat ja astianpesuaineet• Valkaisuainepohjaisia puhdistusaineita (esim. Clorox™*, Sterilox™*)• Vetyperoksidipohjaisia puhdistusaineita• Hankaavia puhdistusaineita (esim. Comet Cleanser™*)• Asetoni- tai hiilivetypohjaisia puhdistusaineita• MEK (metyylietyyliketoni)• Birex®*• Gluteraldehydi• Kvaternaariset ammoniumkloridisuola puhdistusaineita• Cavicide1™*-liuosta tai -pyyhkeitä• Cavicide™*-tuotteet (valkaisuaineettomat)***Muun kuin Ultradent-yhtiön tavaramerkki** Jos käytetään, voi haalistaa värit
AKKULATURI:
Jos puhdistus on tarpeen, irrota laturi, kostuta liina isopropyylialkoholilla, ja pyyhi laturin tai akkujen pinnat. Anna laturin kuivua kokonaan ennen käyttöä.
VALOSUOJAN PUHDISTUS:
95

Tekniset tiedot/Tiedot

Kylmä desinfioi Johdottoman VALO -laitteen valosuoja millä tahansa pintadesinfektioaineella. ÄLÄ käytä autoklaavia.

Komposiitti kovetuksessa toimivat aallonpituuskaistat:

Säilytys ja hävitys

Jos valokovetin varastoidaan yli 2 viikon ajan tai pakataan siirtoa varten, poista aina akut. Jos akut jätetään yksikköön pitkäksi ajaksi ilman uudelleen lataamista, ne voivat menettää
toiminta- tai latauskyvyn. Älä säilytä akkuja yli 60°C:n (140°F:n) lämpötilassa tai suorassa auringonvalossa.

TEHOKKAAT KOMPOSIITTIKOVETETUT AALLONPITUUSKAISTAT(Hyödynnettävä aallonpituusalue: 385-515 nm)
SPEKTRINEN SÄTEILYTEHO (mW/nm)

Valokovettimen säilytys ja kuljetus:• Lämpötila: +10-40°C (+50-104°F)• Suhteellinen kosteus: 10-95 %• Ilmanpaine: 500-1060 hPa
Elektroniikka tuotteita (kuten valokovettimia, latureita, akkuja tai virtaläheitä) hävittäessä, noudata paikkallisia jätteenhävitys ja -kierrätys ohjeita.

Tekniset näkökohdat
Lisävarusteet
Nimike

VALO- suojat

Johdottoman VALO -laitteen
valosuoja

VALO-laturi

Täydennyskoulutustiedot

MDSS GmbH
Schiffgraben 4130175 Hanover
Saksa

Valmistaja:
TIDI Products, LLC.
570 Enterprise Drive
Neenah, WI 54956, USA
Valmistettu USA:ssa

Jälleenmyyjä:
Ultradent Products Inc
505 West Ultradent Drive (10200
South)South Jordan, UT 84095
USA

Tehokas ylempi
aallonpituusalue
440-480 nm

Xtra
Suuri teho
Vakio

WAVELENGTH (nm)Ominaisuus

Tiedot/Määrittelyt

LinssiHalkaisija 11,7 mm

Aallonpi-tuusalue• Käyttökelpoinen aallonpituusalue: 385-515 nm• Huippuaallonpituudet: 395-415 nm ja 440-480 nm
Nimellisen säteilyn eksitanssin vertailukaavio

VALO-akut†* Demetron
L.E.D.
Radiometri† MARC
spektrianalysaattori

Mittarin apertuuri

7 mm

3.9 mmNormaaliteho(±10%)

1000 mW/cm²

Korkea Plus teho(±10%)

1600 mW/cm²

Mittausväline

Valon voimakkuus
talukko

Erittäin korkea teho(±10%)

96

Tehokas alempi
aallonpituuskaista
395-415 nm‡ Gigahertsin spektrianalysaattori
Säteilyneksitanssi

Koko-naisteho

15 mm

15 mm

900 mW/cm²

970 mW

1800 mW/cm2

1500 mW/cm²

1615 mW

3200 mW/cm²(+/-20%)2100 mW/cm²

2260 mW

Säteilyn eksitanssi vaihtelee, riippuen
instrumentin ominaisuuksista, mittaustavasta
ja valon sijoituksesta.
† Demetron-radiometriä ja MARCspektrianalysaattoria tulee käyttää vain
viitteenä, sillä niiden apertuurit on pienempiä
kuin VALO-valokovettimien.
* Demetron-radiometriä tulee käyttää
vain viitteenä sen tehon ja spektrivasteen
rajoituksista johtuen.
‡ Säteilyn eksitanssi on ISO 10650 -standardin
mukainen, kun se mitataan gigahertsin
spektrianalysaattorilla.

VALO Grand-valokovetin

Luokitukset: IEC 60601-1 (turvallisuus), IEC 60601-1-2 (EMC)

Paino:• Akun kanssa: 6 oz. (190g)• Ilman akkua: 5oz. (150g)
Mitat: (8 x 1,28 x 1,06) tuumaa (203 x 32,5 x 27) mm

Laturin virtalähde

Lähtö - 12VDC 500 mA
Tulo - 100-240VAC
Ultradent P/N 5930 VALO-laturin virtalähde kansainvälisillä pistokkeilla

Luokitukset: IEC 60601-1 (turvallisuus)
Johdon pituus - 1,8 metriä (6 jalkaa)
VALO-laturin virtalähde on lääketieteellisen tason, luokka II-virtalähde ja tarjoaa eristeen sähköverkkoa vasten

97

Ominaisuus

Tiedot/Määrittelyt

Ohjeistus ja valmistajan ilmoitus sähkömagneettisista päästöistä
Säteilytesti

Säännösten-mukaisuus

Sähkömagneettinen ympäristö - ohjeistus

RF-säteily
CISPR 11

Ryhmä 1

Valokovetin käyttää akkuja ja siihen ei vaikuta EMI, RF- tai ylijännitesuojaus.

VALO-laturi

VALO 3,6V älykäs litium-rautafosfaattiakkulaturi:• Latauksen automaattinen katkaisu, kun akku on täynnä
• Viallisten akkujen automaattinen tunnistus• Suojaukset: Lämpö, ylilataus, oikosulku, napaisuussuojaus
⸰ Punainen LED – lataa
⸰ Vihreä LED – tyhjä tai täysin latautunut⸰ LED pois päältä – oikosulku• Latausaika: 1-3 tuntia
Luokitukset: CE, WEEE

RF-säteily
CISPR 11

Luokka B

Valokovetin käyttää sähkö- ja sähkömagneettista energiaa vain sisäisiin toimintoihinsa. Sen vuoksi
sen aiheuttama radiotaajuussäteily on erittäin vähäistä, eikä se todennäköisesti aiheuta häiriöitä
lähellä oleviin sähkölaitteisiin.

ES

Ladattava: Turvallisen kemian litium-rautafosfaatti (LiFePO4) RCR123A• Käyttöjännite: 3,2 V
Luokitukset: UL, CE, RoHS, WEEE

Harmoniset päästöt
IEC 61000-3-2

Harmoniset säteily- ja jännittevaihtelujen testaukset eivät ole olennaisia valokovettimen käytössä,koska se käyttää akkuvirtaa.

VALO-akut

Jännitevaihtelut/välkesäteily IEC
61000-3-3

ES

Valokovetin soveltuu käytettäväksi kaikenlaisissa tiloissa, mukaan lukien kotitaloukset sekätilat, jotka on liitetty suoraan julkiseen verkkovirtaan, joka toimittaa matalajänitteistä virtaa
kotitalouskäyttöön.

Lämpötila: +10-32°C (+50-90°F)

Käyttöolosuhteet Suhteellinen kosteus: 10-95 %Ilmanpaine: 700-1060 hPa

Valokovetin on tarkoitettu lyhytaikaiseen käyttöön. Ympäristön enimmäislämpötilassa (+32°C): 1 minuutin yhtämittainen ON-jakso, 30 minuutin
OFF-jakso (jäähdytysjakso).

Käyttöaika

Vianmääritys
Jos alla ehdotetut ratkaisut eivät korjaa ongelmaa, soita Ultradentille nroon +1-800.552.5512. Yhdysvaltojen ulkopuolella, soita Ultradentin tukkumyyjälle taihammaslääkärien jälleenmyyjälle.
Ongelma

Mahdolliset ratkaisut

Valo ei käynnisty

1) Paina joko Aika/Tila-vaihtopainiketta tai virtapainiketta virransäästötilasta heräämiseen.
2) Akun varaustilan selvittämiseksi tarkista punainen Alhainen akun varaustaso -ilmaisin.
3) Tarkista, että uudet akut on asennettu laitteeseen oikein.
4) Jos punainen ja keltainen varoitusmerkkivalo vilkkuu, se tarkoittaa että valokovetin on saavuttanut sen sisälämpötilan
turvarajan. Anna valokovettimen jäähtyä 10 minuuttia tai käytä kosteaa viileää pyyhettä yksikön jäähdyttämiseen
nopeasti.
5) Jos punainen varoitusmerkkivalo vilkkuu ja piippaa jatkuvasti, soita Ultradentin asiakaspalveluun korjausta varten.

Valo ei pysy päällä haluttua ajanjaksoa

1) Tarkista Tilan ja ajastuksen valoista, että syötettynä on oikea aika.
2) Akun varaustilan selvittämiseksi tarkista Alhainen akun varaustaso -ilmaisin.
3) Tarkista, että uudet akut on asennettu laitteeseen oikein.

Valo ei koveta aineita kunnolla

1) Tarkista, ettei linssissä ole kovetetun hartsin/komposiitin jäämiä.
2) Käytä asianmukaisia meripihkan värisiä UV-suojalaseja, ja tarkista, että LED-valot toimivat.
3) Tarkista tehon taso valomittarilla. Jos käytät valomittaria, Ultradent suosittelee valokovettimen tarkastamista Normaaliteho -tilassa.
HUOMAUTUS: Todellinen numeerinen teho vääristyy yleisten valomittareiden epätärkkuuden ja valokovettimen käyttämän
erikoisvalmisteisen LED-pakkauksen johdosta. Valomittarit poikkeavat toisistaan merkittävästi, ja ne on suunniteltu tietyillevalonohjauskärjille ja linsseille.
4) Tarkista aineen viimeinen käyttöpäivä.
5) Varmista, että käytetään valmistajan suosittelemaa asianmukaista tekniikkaa.

Akut eivät lataudu

1) Varmista, että akut on asetettu laturiin oikein päin ja anna niiden latautua 1-3 tuntia.
2) Jos laturin punaiset valot eivät muutu vihreäksi, soita Ultradentin asiakaspalveluun ja tilaa uudet akut ja/tai laturi.
3) Jos kumpikaan vihreästä tai punaisesta laturin valosta ei ole näkyvissä, soita Ultradentin asiakaspalveluun ja tilaa uusi
laturi ja/tai verkkolaite.

Laturi ei lataa akkuja

1) Varmista, että laturi on kytketty laitteeseen ja verkkolaite toimivaan pistorasiaan.
2) Jos kumpikaan vihreästä tai punaisesta laturin valosta ei ole näkyvissä, soita Ultradentin asiakaspalveluun ja tilaa uusi
laturi ja/tai verkkolaite.

Tilaa tai aikavälejä ei voi muuttaa

1) Pidä sekä Aika/Tila- että virtapainiketta alhaalla, kunnes kuuluu sarja piippauksia osoittaen, että valokovetin on auki.

Ohjeistus ja valmistajan ilmoitus sähkömagneettisesta häiriönsiedosta
Valokovetin on tarkoitettu käytettäväksi alla määritellyssä sähkömagneettisessa ympäristössä. Asiakkaan tai käyttäjän tulisi varmistaa, että sitä käytetään tällaisessa
ympäristössä.
HÄIRIONSIETO-TESTIIEC 60601 -testitaso

Säännöstenmukaisuus-taso

Sähkömagneettista ympäristöä koskeva ohjeistus

Sähköstaattinen purkaus(electrostatic discharge, ESD)

±8 kV kontaktipurkaus±15 kV ilmapurkaus±8 kV kontaktipurkaus±15 kV ilmapurkaus

Fyysistä ympäristöä tulisi rajoittaa seuraavasti:
1) IP-koodi: IP202) Älä upota nesteeseen.
3) Älä käytä syttyvien kaasujen läheisyydessä.
Yksiköllä on ei-APG ja ei-AP -luokitus.
4) Kosteusrajat säilytyksessä: 10-95 %5) Säilytyslämpötilat: +10-40°C

Nopea sähköinen transientti/
purskeIEC 61000-4-4±2 kV virransyöttölinjoissa±1 kV tulo-/lähtölinjoissa

Ylijännite
IEC 61000-4-5±1 kV linjalta linjalle±2 kV linjalta maahan±2 kV virransyöttölinjoissa
VALO GRAND on akkukäyttöinen ja sitä ei voi yhdistää
Huomautus 1: Valokovettimessa ei sähköverkkoon.
ole tulo-/lähtöporttia (I/0)
Koska valokovetin on akkukäyttöinen, sähköiset
transientit, sähköiskut, jännitekuopat, oikosulut,±1 kV linjalta linjalle
jakelun keskeytykset tai sähköverkon vaihtelut eivät±2 kV linjalta maahan
vaikuta siihen.

IEC 61000-4-2

Voltage, dips, shorts, interruptions <5% U
and variations on the power(>95%:n lasku U:ssa 0,5 syklin
supply input lines
aikana)<5% U(>95%:n lasku U:ssa 0,5 syklin
aikana)Edellä mainitut voivat vaikuttaa lisävarusteisiin
kuuluvaan akkulaturiin, mutta sen toiminta on
erillistä eikä siten kriittinen Johdottoman VALO
-valokovettimen kannalta.

IEC 61000-4-11

40% U
40% U
(60%:n lasku U:ssa 5 syklin aikana) (60%:n lasku U:ssa 5 syklin aikana) Jos valokovettimen akun jännite laskee alle 4V, laite
estää toiminnan. Johdoton VALO -valokovetin kytkeytyy
70% U
70% U
pois päältä. Kun uudet akut on laitettu paikoilleen(30%:n lasku U:ssa 25 syklin(30%:n lasku U:ssa 25 syklin
ja asianmukainen tehotaso palautuu, valokovetin
aikana)aikana)käynnistyy uudelleen ja palaa ennen tehon menetystä
olevaan tilaan. Valokovetin palaa toimintaan
<5% U
<5% U
automaattisesti virran menetyksen päätyttyä.
(>95%:n lasku U:ssa 5s aikana)(>95%:n lasku U:ssa 5s aikana)
Huomautus 2: Automaattinen
toimintaan paluu

Virran taajuus(50/60 Hz) magneettikent-tä

30 A/m30 A/mIEC 61000-4-8

Virran taajuus magneettikenttien tulisi
vastata tasoltaan tyypillisen paikan, kuten
kotitalous-, kotihoito-, liikeyritys-, sairaala- taivaruskuntaympäristön tyypillistä tasoa.

Huomautus 1: Valokovetinta ei ole varusteltu tulo-/lähtöporteilla, eikä näkyvissä olevilla tulo-/lähtölinjoilla.

Lisätiedot
Ohjeistus ja valmistajan ilmoitus sähkömagneettisista päästöistä
Valokovetin on tarkoitettu käytettäväksi alla määritellyssä sähkömagneettisessa ympäristössä. Asiakkaan tai käyttäjän tulisi varmistaa, että sitä käytetään tällaisessa
ympäristössä.
98

99

Ohjeistus ja valmistajan ilmoitus sähkömagneettisesta häiriönsiedosta muita kuin elämää ylläpitäviä järjestelmiä koskien
Valokovetin on tarkoitettu käytettäväksi alla määritellyssä sähkömagneettisessa ympäristössä. Asiakkaan tai käyttäjän tulisi varmistaa, että sitä käytetään tällaisessa
ympäristössä.

HÄIRIONSIETOTESTI IEC 60601 -testitaso

Säännöstenmukaisuustaso

Johtunut
radiotaajuussä-teily

3 Vrms3 VrmsIEC 61000-4-6

150 kHz–80 MHz

150 kHz–80 MHz

Sähkömagneettista ympäristöä koskeva ohjeistus
Kannettavia ja liikuteltavia radiotaajuusviestintälaitteita ei tulisi käyttää lähempänä mitään
valokovettimen osaa, mukaan lukien kaapelit, kuin suositelluksi suojaetäisyydeksi onlaskettu lähettimen taajuuteen sovellettavaa yhtälöä käyttäen.
Suositeltu suojaetäisyys:

80-800 MHz

800 MHz-2,5 GHzRadiotaajuusä-teily

3 V/m

3 V/m

P on lähettimen valmistajan ilmoittama lähettimen nimellinen enimmäissyöttöteho
watteina (W) ja d on suositeltava suojaetäisyys metreinä (m).
Kiinteiden RF-lähettimien kenttävoimakkuus, kuten määritettynä sähkömagneettisessa
kartoituksessaa, tulisi olla alhaisempi kuin säännöstenmukaisuustaso kullakintaajuusalueellab vaatii.

IEC 61000-4-3

80 MHz–2,5 GHz

80 MHz–2,5 GHz

Seuraavalla symbolilla merkittyjen laitteiden läheisyydessä saattaa esiintyä häiriöitä:Ohjeistus ja valmistajan ilmoitus suositellusta suojaetäisyydestä kannettavien ja liikuteltavien radiotaajuusviestintälaitteiden ja Johdottoman VALO -valokovettimen
välillä.
Valokovetin on tarkoitettu käytettäväksi sähkömagneettisessa ympäristössä, jonka säteileviä radiotaajuushäiriöitä valvotaan. Valokovettimen käyttäjä voi auttaa
estämään sähkömagneettisia häiriöitä säilyttämällä alla suositellut vähimmäisetäisyydet kannettavien ja liikuteltavien radiotaajuutta käyttävien viestintälaitteiden(lähettimien) ja valovettimen välillä. Suositukset perustuvat tietoliikennelaitteiden maksimitehoon.
Lähettimen taajuuden vaatima suojaetäisyys(metriä)Nimellinen enimmäisantoteho
lähettimelle(P watteina)

150 kHz – 80 MHz

80 MHz – 800 MHz

800 MHz – 2,5 GHz

0,01

0,12 metriä

0,035 metriä

0,07 metriä

0,10,37 metriä

0,11 metriä

0,22 metriä

1

1,7 metriä

0,35 metriä

0,7 metriä

10

3,7 metriä

1,11 metriä

2,22 metriä

100

11,7 metriä

3,5 metriä

7,0 metriä

Valokovetin on testattu IEC 60601-1-2:2014 standardin mukaisesti ja se läpäisi säteilykentänteho testin seuraavilla voimakkuusarvoilla: 10V/m, 80 kHz:n-2,5 GHz:ntaajuusalueella. Arvo 3Vrms viittaa edellä mainitun kaavan arvoon V1 ja arvo 10V/m viittaa kaavan arvoon E1.
Jos lähettimen suurinta nimellislähtötehoa ei ole mainittu edellä, suositeltu etäisyys d metreinä (m) voidaan arvioida lähettimen taajuuteen sovellettavalla yhtälöllä,jossa P on valmistajan ilmoittama lähettimen suurin nimellislähtöteho watteina (W).
HUOMAUTUS 1: 80 MHz:n ja 800 MHz:n taajuusalueella pätee korkeamman taajuusalueen suojaetäisyys.
HUOMAUTUS 2: Nämä ohjeet eivät välttämättä sovellu kaikkiin tilanteisiin. Imeytyminen ja heijastuminen rakenteista, esineistä ja ihmisistä vaikuttaa
sähkömagneettisten aaltojen etenemiseen.

HUOMAUTUS 1: 80 MHz:n ja 800 MHz:n taajuusalueella sovelletaan korkeampaa taajuusaluetta.
HUOMAUTUS 2: Nämä ohjeet eivät välttämättä sovellu kaikkiin tilanteisiin. Imeytyminen ja heijastuminen rakenteista, esineistä ja ihmisistä vaikuttaa
sähkömagneettisten aaltojen etenemiseen.
aKiinteiden lähettimien, kuten radion tukiasemien (matka-/langattomien) puhelimien ja matkaviestinverkon toistimien, amatööriradioiden, AM- ja FMradiolähetysten ja TV-lähetysten kentän voimakkuuksia ei voida ennustaa teoreettisesti kovin tarkasti. Kiinteiden radiolähettimien aiheuttaman säteilyn vuoksi,
kannattaisi harkita ympäröivän alueen sähkömagneettista kartoitusta. Jos valokovettimen käyttösijannissa mitattu kentänvoimakkuus ylittää sitä koskevan
radiotaajuuden säädöstenmukaisuustason, Johdotonta VALO -valokovetinta tulisi tarkkailla normaalin toiminnan varmistamiseksi. Jos havaitaan epänormaalia
toimintaa, lisätoimenpiteet, kuten valokovettimen uudelleen suuntaaminen tai sijoittaminen, saattavat olla tarpeen.
b
150 kHz:n-80 MHz:n ylittävällä taajuusalueella kenttävoimakkuuksien tulisi olla alle 3 V/m.

100

101

VALO™ GRANDNOHERDINGSLYS

Produktbeskrivelse

Med sitt bredbåndsspekter er VALO Grand designet for å polymerisere alle lysherdede produkter i bølgelengdeområdet 385-515 nm per ISO 10650. VALO Grand-herdingslys bruker
Ultradent VALO oppladbare batterier og batterilader. Herdingslyset er utformet for å hvile i en standard dentalenhetsbrakett eller kan monteres ved hjelp av braketten som følger med
settet.

VALO Grand-produktkomponenter:
•
•
•
•
•
•

Merk:

• Ved å bruke den hygieniske barrierehylsen reduseres lyset med 5-10 %. På grunn av den høye effekten av herdingslyset, har herding vist seg å være vesentlig ekvivalent.
• Herdingslyset må rengjøres og saniteres med passende rengjørings- og/eller sanitiseringsmidler etter hver pasient. Se avsnittet Behandling.

VALO trådløs lysskjerm:
Knapper og lys

Brosjyre

På/av-knapp(er)(Toppen eller bunnen)
Tidtakerlys
Lavt batteri – Indikator
Modus-/statuslys
Endre-knapp for tidtaker/
modus

VALO trådløs lysskjerm er ovalformet, kan roteres for maksimal bruk, og kan brukes med gjennomsiktig barrierehylse.

Bruk1) Hver effekt-modus brukes til herding av dentalmaterialer med fotoinitiatorer. Se Hurtigmodusveiledning for anbefalte herdetider.
MERK: Herdingslyset er programmert til å sykle fra Standard effekt- til Høy effekt pluss-, til Ekstra effekt-modus i sekvens. For eksempel, for å bytte fra Standard effekt-modus til Ekstra
effekt-modus, er det nødvendig å veksle fra Høy effekt pluss-modus og deretter til Ekstra effekt-modus.
2) Herdingslyset lagrer det sist brukte tidsintervallet og modusen, og det kommer tilbake til dette når modiene endres eller hvis batteriene fjernes.

Drift
HERDINGSMODUS: Standard effekt-modus

TIDSINTERVALLER: 5, 10, 15, 20 sekunder.
• Herdingslyset er standard i denne modusen når det FØRST er slått på. Modus-/statuslyset blir grønt og de fire grønne tidtakerlampene lyser, som indikerer Standard effekt-modus.
• For å endre tidsintervaller, trykker du raskt på tid-/modusknappen.
• Trykk på strømknappen for å herde. For å stoppe herdingen før ferdigstillelse av et tidsintervall, trykk på strømknappen igjen

HERDINGSMODUS: Høy strøm pluss-modus(Topp)For alle produkter som er beskrevet heri, les nøye og forstå all instruksjon og SDS-informasjon før bruk.

Indikasjoner for bruk / tiltenkt formål

Kilden til belysning for herding av fotoaktiverte dentalreparative materialer og klebemidler.

Advarsler og forholdsregler
Risikogruppe 2
FORSIKTIG UV-utslipp fra dette produktet. Øye- eller hudirritasjon kan skyldes eksponering. Bruk passende skjerming.
FORSIKTIG Mulig farlig optisk stråling utstrålet fra dette produktet. Ikke stirr på driftslampen. Kan være skadelig for øynene.
• Ikke se direkte inn i lysutgangen. Pasient, kliniker og assistenter skal alltid bruke gulbrun UV-beskyttelse når herdingslys er i bruk.
• For å unngå fare for elektrisk støt, er det ikke tillatt å endre utstyret. Bruk bare den/de medfølgende Ultradent VALO-strømforsyningen og -pluggadapterne. Hvis disse komponentene
er skadet, må du ikke bruke dem og ringe Ultradent kundeservice for å bestille en erstatning.
• Bærbart RF-kommunikasjonsutstyr kan forringe ytelsen hvis det brukes nærmere enn 30 cm (12 tommer).
• Bruk kun autorisert tilbehør, kabler og strømforsyninger for å forhindre feilaktig bruk, økte elektromagnetiske utslipp eller redusert elektromagnetisk immunitet (se avsnittet
Elektromagnetisk utslipp).
• For å unngå fare for elektrisk brann i forbindelse med håndtering av batterier:⸰ IKKE bland oppladbare batterier med ikke-oppladbare batterier eller andre batterityper.
⸰ IKKE forsøk å lade ikke-oppladbare batterier.
⸰ IKKE autoklaver eller spray batterier, batterikontakter, lader eller strømadapter med væske av noe slag. Hvis det oppstår korrosjon på kontaktene til batteriladeren, ring Ultralentkundeservice for å bestille en erstatning.
⸰ IKKE lad batterier rundt brannfarlige materialer.
⸰ IKKE hold laderen i klinisk operasjon.
• For å unngå fare for skade, bruk IKKE batterier som er korrodert (rust), bulket, avgir lukt eller væsker, har revnet eller mangler innpakning eller ellers skadet. Ring til Ultradentkundeservice for å bestille erstatningsbatterier.
• For å unngå fare for termisk irritasjon eller skade, unngå bakside-mot-bakside-herding og ikke utsett munnvev i umiddelbar nærhet i mer enn 10 sekunder i enhver modus. Hvis det
kreves lengre herdetider, bruk flere kortere herdesykluser eller bruk et dobbeltherdeprodukt for å unngå oppvarming av mykt vev.
• Vær forsiktig når du behandler pasienter som lider av uønskede fotobiologiske reaksjoner eller følsomheter, pasienter som behandles med kjemoterapi eller pasienter som
behandles med fotosensibiliserende medisiner.
• Denne enheten kan være utsatt for sterke magnetiske eller statiske elektriske felter, noe som kan forstyrre programmeringen. Hvis du mistenker at dette har skjedd, trekker du ut
støpselet midlertidig og kobler det deretter til stikkontakten.
• Tørk ikke av herdingslyset med kaustiske eller slipende rengjøringsmidler, autoklaver eller nedsenk i noen form for ultralydsbad, desinfeksjonsmiddel, rengjøringsmiddel eller væske.
Unnlatelse av å følge medfølgende behandlingsinstruksjoner kan gjøre herdingslyset ubrukelig.
• For å unngå skade på utstyret, IKKE sett inn fingre, instrumenter eller andre gjenstander i batterirommet i herdingslyset.
• For å unngå skade på utstyret, prøv IKKE å rengjøre gullkontaktene eller deler av batterirommet. Ring til Ultradent kundeservice hvis du har noen spørsmål.
1021) Lad batteriene før du bruker herdingslyset (se avsnittet Vedlikehold av batteriet).
2) Plasser herdingslyset i en standard dentalmonteringsbrakett eller monteringsbrakett for tilbehør til klar for bruk.
3) Før hver bruk legges en ny barrierehylse over herdingslyset..
Den hygieniske barrierehylsen er tilpasset herdingslyset og holder overflaten på herdingslyset ren. Barrierehylsen bidrar til å hindre krysskontaminering, bidrar til attannkomposittmaterialet ikke kleber seg fast på overflaten av linsen og herdingslyset, og forhindrer misfarging og korrosjon fra rengjøringsløsninger.

Bilde(Bunn)

Trinnvise instruksjoner
Forberedelse

Installere hygieniske barrierehylser:

1 - VALO stort herdingslys
4 - Ultradent VALO oppladbare batterier
1 - Ultradent VALO-batterilader med medisinsk 12 VDC strømadapter
1 - VALO barrierehylse-prøvepakke
1 - VALO stor lysskjerm
1 - Herding av lysoverflatemonteringsbrakett med dobbelt klistrebånd

Oversikt over kontroller:• For å forhindre risikoen for krysskontaminering, er barrierehylser til engangsbruk.
• For å redusere risikoen for korrosjon, fjern barrierehylsen etter bruk.
• For å redusere risikoen for underherdede harpikser, bruk ikke herdingslys hvis linsen er skadet.

TIDSINTERVALLER: 1, 2, 3, 4 sekunder.
• Fra Standard effekt-modus, trykk og hold inne tids-/modusendringsknappen i 2 sekunder og slipp opp. Modus-/statuslyset vil være oransje, og de fire grønne tidtakerlampene villyse og blinke, som indikerer Høy effekt-modus.
• For å endre tidsintervaller, trykker du raskt på tid-/modusknappen.
• Trykk på en av på/av-knappene for å herde. For å stoppe herdingen før tidsintervallet er gjennomført, trykker du på en av på/av-knappene igjen.
• For å gå tilbake til Standard effekt-modus, trykk og hold inne tids-/endringsknappen i 2 sekunder og slipp, dette vil veksle til Ekstra effekt-modus. Trykk og hold igjen i 2 sekunder,og slipp. Modus-/statuslyset blir grønt og de fire grønne tidtakerlampene lyser, som indikerer standardmodus.

HERDINGSMODUS: Ekstra effekt-modus

TIDSNTERVALL: Bare 3 sekunder (Merk: Xtra effekt-modusen har en 2 sekunders sikkerhetsforsinkelse på slutten av hver herdingsperiode for å begrense oppvarming under konsekvent
herding. På slutten av forsinkelsen indikerer piping at enheten er klar til fortsatt bruk).
• Fra Standard effekt-modus, trykk på tids-/modusendringsknappen i 2 sekunder, slipp, trykk og hold igjen i 2 sekunder, og slipp opp. Modus-/statuslyset vil bli oransje og blinke, og
tre av de grønne tidtakerlampene vil lyse og blinke, som indikerer Ekstra effekt-modus.
• Trykk på strømknappen for å herde. For å stoppe herdingen før ferdigstillelse av et tidsintervall, trykk på strømknappen igjen.
• For å gå tilbake til standard strømmodus, trykk og hold inne tids-/modusknappen i 2 sekunder og slipp opp. Modus-/statuslyset lyser grønt, og de grønne tidtakerlampene er tent,
noe som indikerer standard strømmodus.

HVILE-modus: Herdingslyset går inn i hvilemodus etter 60 sekunder med inaktivitet, som indikert ved langsom blinking av modus-/statuslys. Når du henter opp eller berører enheten, vil
det vekke herdingslyset og automatisk returnere det til den sist brukte innstillingen. For å maksimere batteriets levetid, la herdingslyset være uforstyrret når det ikke er i bruk.

Rens

1) Kast brukte barrierehylser i vanlig avfall etter hver pasient.
2) Se avsnittet Behandling.

Monteringsbrakettinstruksjoner

1) Braketten skal monteres på en flat, oljefri overflate.
2) Rengjør overflaten med desinfeksjonssprit.
3) Trekk av brakettens tape.
4) Plasser braketten slik at herdelyset løftes oppover når det fjernes. Trykk godt på plass.

103

Hurtigmodusveiledning:Standard effekt
1000 mW/cm2

Modus

Høy effekt pluss
1600 mW/cm2

Ekstra effekt
3200 mW/cm2

På/Av-knapp
Modus-/tidtakning-LED-lamper

Tidsknapper

Tidsalternativer

5s

10s

15s

20s

1s

2s

3s

4s

Slik endrer du tiden

Trykk og slipp tidsknappen raskt for å gå gjennom tidsalternativene.

Slik endrer du modi

Trykk og hold inne tasten i 2 sekunder og slipp opp. Herdingslyset vil veksle til neste modus.

Brosjyre

Kontinuerlige LED-lamper

Kun 3sBlinkende LED-lamper

Svake batterier: Herdingslyset signaliserer brukeren at det er på tide å bytte batterier når lavt batteri-indikatorlyset blinker rødt. Hvis batteriladingen blir for lav, vil det høres et lydsignal
med 3 pip og herdingslyset tillater ikke videre drift før batteriene er ladet opp eller nye batterier er satt inn. (Se Hurtigveiledning)

Curing Mode

Power* (mW)

Irradiance* (mW/cm2)

Total Exposure Time (Seconds)

Energy** (Joules)

Standard

970

900

5/10/15/20

4.8J/9.7J/14.5J/19.4J

High Power Plus

1,620

1,500

1/2/3/4

1.6J/3.2J/4.8J/6.5J

Xtra

2,260

2,100

3

6.8J

Bytt ut batterier

Ladetid og batterilevetid: Fulladet batterilevetid i herdingslyset er avhengig av modus-/tidsintervall, batteritype, brukstid og LED-effektivitet. Generelt bør oppladbare batterier vare 1-2
uker. Ikke oppladbare batterier kan vare 2-3 ganger lenger.
• Anbefalt oppladingsintervall: Når indikatoren for lavt batteri lyser, eller omtrent hver 1-2 uker, avhengig av bruk.
• Ekstra batterier: Herdingslyset leveres med 4 oppladbare batterier. Vi anbefaler også å ha et ekstra sett med ikke-oppladbare CR123A-batterier tilgjengelig hvis det oppstår
problemer eller tap av de oppladbare batteriene.
• Batterilevetid: Oppladbare litiumjernfosfatbatterier kan lades opp omtrent 1000 til 2000 ganger. Basert på normal bruk og riktig vedlikehold, bør batteriene vare opptil fem år, menskal byttes etter behov.

Garanti

Hurtigadvarselsveiledning:• Lavt batteri: sakte blinkende• Slå av: 3 pip, blinkende• Forbyr drift

Slik bytter / setter du inn batterier:
1) Fjern bakdekselet ved å dreie det en kvart omdreining mot klokken.
2) Fjern batteriene.
3) Sett inn nye batterier med den positive (+) enden først.
4) Sett bakdekselet på plass ved å justere og forsiktig skyve mens du dreier det med klokken. Dekselet klikker når det sitter riktig på plass.
5) Enheten er klar til bruk.
Ultradent autoriserer om nødvendig følgende IKKE-OPPLADBARE batterier for herdingslyset:
MERK: IKKE forsøk å lade ikke-oppladbare batterier.
1) Tenergy Propel Photo Lithium2) Titanium Innovations CR123A
3) Energizer® 1234) Duracell® Ultra CR123A
5) SureFire® SF123A
6) Panasonic® CR123A

Hurtigherdingsveiledning:

Effektnivå-advarsel

Slik lader du batterier:
1) Koble laderen til stikkontakten.
2) Sett batteriene i laderen med den positive (+) enden rettet mot indikatorlysene på laderen.
3) Grønt lys indikerer at batteriene er klare til bruk.
4) Det vil ta 1–3 timer å lade batteriene. La batteriene være i laderen til de er klare til bruk.
MERK: Hvis det røde lyset på laderen ikke blir grønt når batteriene har ladet lenger enn tre timer, kan batteriet bli ødelagt (ødelagt) og kan ikke lades. Prøv med et nytt batteri, eller ringtil Ultradent kundeservice for å bestille et nytt sett med oppladbare batterier.

Temperatur-advarsel
Tillat for avkjøling
• 3 pip• Sakte blinking• Forbyr drift

Service-advarsel

LED-advarsel

Ring kundeservice for reparasjon

Ring kundeservice for reparasjon• Ingen lyd• Blinker hvert. 2 sekund• Tillater drift• 3 sammenhengende pip• Enkel hurtigblits• Forbyr drift

Ultradent garanterer herved at dette instrumentet i en periode på 5 år* skal være i samsvar med alle spesifikke krav til spesifikasjonene som beskrevet i Ultradents dokumentasjon som
følger med produktet, og være fritt for eventuelle mangler i materialer eller utførelse. Denne garantien gjelder kun for den opprinnelige kjøperen og kan ikke overføres. Alle defekte
produkter skal returneres til Ultradent. Det finnes ingen brukerkomponenter i VALO Cordless-herdingslyssystemet. Tukling med VALO Cordless-herdingslys vil ugyldiggjøre garantien.
VALO Cordless-herdingslysets garanti dekker ikke kundeskader. For eksempel; Hvis en VALO Cordless blir misbrukt eller tapt og linsen går i stykker, vil kunden være ansvarlig for å betalefor eventuelle nødvendige reparasjoner.
*Med salgskvittering som angir salgsdatoen til tannlegen.

Behandling

Etter hver bruk skal overflaten og linsen tørkes av med et gasbind eller en myk klut fuktet med et godkjent overflatedesinfeksjonsmiddel.
GODKJENTE RENGJØRINGSMIDLER:• Lysol Brand III desinfeksjonsspray (Anbefalt)• Isopropylalkohol• Etylalkoholbaserte rengjøringsmidler• Lysol®*-konsentrat (kun alkoholbasert)

Vedlikehold
Reparasjon

Bruker-utført reparasjon
1) Kontroller linsen jevnlig for herdede dentalharpikser. Bruk et dentalinstrument som ikke er laget av diamant for å forsiktig fjerne eventuell herdet harpiks.
2) Lysmålere varierer sterkt, og er designet for spesifikke lysledningstupper og -linser. Ultradent anbefaler rutinemessig å sjekke utgangen i Standard effekt-modusen. MERK: Den sannenumeriske utgangen vil være forskjøvet på grunn av unøyaktigheten til vanlige lysmålere og den egendefinerte LED-pakken i herdingslyset.
Produsentreparasjon
1) Reparasjoner skal kun utføres av autorisert servicepersonell. Ultradent gir servicepersonell dokumentasjon for å utføre reparasjoner.
2) Når du sender enheter til reparasjon, service eller kalibrering, må du alltid fjerne batteriene fra herdingslyset og laderen. Sett batteriene, laderen, adapteren og herdingslyset separati returboksen.
3) Send batterier i henhold til lokale forskrifter.

UAKSEPTABLE RENSEMIDLER - IKKE BRUK:• Sterkt alkalisk vaskemiddel av enhver type, inkludert håndsåper og oppvasksåper
• Blekemiddelsbaserte rengjøringsmidler (for eksempel Clorox™*, Sterilox™*)• Hydrogenperoksidbaserte rengjøringsmidler• Slipemidler (f.eks. Comet Cleanser™*)• Aceton- eller hydrokarbonbaserte rengjøringsmidler• MEK (metyletylketon)• Birex®*• Gluteraldehyd• Kvaternære ammoniumkloridsaltbaserte rengjøringsmidler• Cavicide1™* løsning eller kluter• Cavicide™*-podukter (ikke-blekemiddel)***Varemerket til et annet selskap enn Ultradent**Hvis det brukes, kan fargen falme

Batterivedlikehold

Lade og bytte batterier
Herdingslyset leveres med 4 oppladbare litiumjernfosfatbatterier.

104

105

BATTERILADER:
Hvis det blir nødvendig å rengjøre det, må du koble fra laderen, fukte en klut med isopropylalkohol, og tørk forsiktig av laderen eller batteriene. La laderen tørke helt før du fortsetter
bruken.

Teknisk informasjon/data

Effektiv komposittherding-bølgelengdebånd:RENGJØRING AV LYSSKJERMEN:
Kalddesinfiser VALO Cordless-lysskjerm ved hjelp av et desinfeksjonsmiddel. IKKE autoklaverer.

EFFEKTIVE KOMPOSITT HERDENDE BØLGELENGDE BÅND(Utnyttbart bølgelengdeområde: 385–515 nm)

Oppbevaring og kassering

SPEKTRAL STRÅLEKRAFT (mW/nm)

Hvis herdingslyset lagres i perioder som er lengre enn 2 uker, eller pakker det for reise, må du alltid fjerne batteriene. Hvis batteriene etterlates i enheten i lange perioder uten
oppladning, kan det hende at de slutter å fungere eller ikke lar seg lade opp igjen. Ikke lagre batteriene i temperaturer over 60 °C eller i direkte sollys.
Lagring og transport av herdingslyset:• Temperatur: +10 °C til +40 °C (+50 °F til +104 °F)• Relativ luftfuktighet: 10 % til 95 %• Omgivelsestrykk: 500 hPa til 1060 hPa
Ved kassering av elektronisk avfall (dvs. herdelys, ladere, batterier og strømforsyninger), følg lokale retningslinjer for avfall og resirkulering.

Tekniske hensyn
Tilbehør
Artikkel

VALO barrierehylser

VALO trådløs lysskjerm

CE-informasjon

MDSS GmbH
Schiffgraben 4130175 HanoverTyskland

Produsert av:
TIDI Products, LLC.
570 Enterprise Drive
Neenah, WI 54956
Laget i USA

Distribuert av:
Ultradent Products Inc
505 West Ultradent Drive (10200
South)South Jordan, UT 84095
USA

Effektiv lavere
bølgelengdebånd
395-415 nm

Effektivt øvre bølgelengdebånd
440-480 nm

Xtra
Høy kraft
StandardBØLGE ENGTH (nm)Egenskap

Informasjon/spesifikasjon

inse

Diameter 11,7 mm

Bølgelengdeområde

• Brukbart bølgelengdeområde: 385–515 nm• Topp for bølgelengder: 395–415 nm og 440–480 nm
Sammenligningsdiagram for nominell utstrålingstetthet

VALO-laderMåle-instrument†* Demetron
L.E.D.
Radiometer† MARC
spektrumanalysatorÅpning av måler

7 mm

3.9 mmStandard effekt(±10%)

1000 mW/cm²

Høy effekt pluss(±10%)

1600 mW/cm²

VALO-batterier
Lysinten-sitetstabell

Ekstra effekt (±10%)‡ Gigahertz spektrumanalysator
TetthetTotal effekt

15 mm

15 mm

900 mW/cm²

970 mW

1800 mW/cm2

1500 mW/cm²

1615 mW

3200 mW/cm²(+/-20%)2100 mW/cm²

VALO stort herdelys Klassifiseringer: IEC 60601-1 (Sikkerhet), IEC 60601-1-2 (EMC)

Laderens strømforsyning

106

Utgang - 12 VDC ved 500mA
Inngang - 100 VAC til 240 VAC
Ultradent delenummer 5930 VALO-laderens strømforsyning med
internasjonale plugginnsatser

Utstrålingstetthet vil variere basert på
instrumentkapabilitet, målingsmetode og
lysplassering.
† Demetron radiometre og MARC
spektrumanalysatorer bør kun brukes som
referanse, da de har mindre åpninger enn
VALO herdelys.
* Demetron radiometre bør kun brukes som
referanse grunnet begrensninger i effekt og
spektralsignatur.
‡ Utstrålingstetthet samsvarer med
ISO 10650 når målt med en Gigahertz
spektrumanalysator.

2260 mW
Vekt:• Med batterier: 6 oz. (190 gram)• Uten batterier: 5 oz. (150 gram)
Dimensjon: (8 x 1,28 x 1,06) tommer, (203 x 32,5 x 27) mm
Klassifiseringer: IEC 60601-1 (Sikkerhet)
Ledningslengde - 6 fot (1,8 meter)
VALO-laderens strømforsyning er en medisinsk klasse IIstrømforsyning og gir isolasjon fra strømnettet

107Diverse informasjon

Egenskap

Informasjon/spesifikasjon

Veiledning og produsentens erklæring om elektromagnetiske utslipp

VALO-lader

VALO 3.6VDC Lithium Iron Phosphate smart battery charger:• Automatisk avstenging når den er fulladet• Automatisk deteksjon av defekte batterier• Beskyttelser: Termisk, overbelastning, kortslutning, omvendt polaritet
⸰ Rød LED – Lader⸰ Grønn LED – Tomt eller fulladet⸰ LED av – kortslutning• Ladetid: 1–3 timer
Klassifisering: CE, WEEE

VALO-batterier

Oppladbar: Sikker kjemi, litiumjernfosfat (LiFePO4) RCR123A• Arbeidspenning: 3,2 VDC
Klassifiseringer: UL, CE, RoHS, WEEE

Driftsforhold

Temperatur: +10 °C til +32 °C (+50 °F til +90 °F)
Relativ luftfuktighet: 10 % til 95 %Omgivelsestrykk: 700 hPa til 1060 hPa

Driftssyklus

Herdingslyset er konstruert for kortsiktig drift. Ved maksimal omgivelsestemperatur (32 °C): 1 minutt PÅ i påfølgende syklus, 30 minutter AV(avkjølingsperiode).

Herdingslyset er beregnet for bruk i det elektromagnetiske miljøet som er angitt nedenfor. Kunden eller brukeren må sørge for at den brukes i et slikt miljø.

Feilsøking

Utslippstest

SamsvarElektromagnetisk miljø - veiledning

RF-utslipp
CISPR 11

Gruppe 1

Herdingslyset bruker batterier og påvirkes ikke av EMI, RF eller overspenning.

RF-utslipp
CISPR 11

Klasse B

Herdingslyset bruker kun elektrisk og elektromagnetisk energi for dets interne funksjoner. Derfor
er eventuelle RF-utslipp svært lave, og vil ikke forårsake forstyrrelser i nærliggende elektronisk
utstyr.

Harmoniske utslipp
IEC 61000-3-2

Ikke aktuelt

Harmonisk utslipp og spenningsfluktuasjonstesting gjelder ikke for herdingslyset fordi det er
batteridrevet.

Spenningsfluktuasjoner/
flimmerutslipp IEC 61000-3-3

Ikke aktuelt

Herdingslyset er egnet for bruk i alle virksomheter, inkludert husholdninger og de som er direkte
forbundet med det offentlige lavspente strømforsyningsnettverket som leverer bygninger til
husholdningsbruk.

Veiledning og produsentens erklæring om elektromagnetisk immunitet

Hvis løsningene som foreslås nedenfor ikke løser problemet, kan du ringe til Ultradent på 800.552.5512. Utenfor USA kan du ringe til din leverandør av Ultradent eller
tannlegeutstyr.
ProblemMulige løsninger

Lyset vil ikke slå seg på1) Trykk på tid-/modus- eller effekt-knappen for å våkne fra strømsparingsmodus.
2) Kontroller den røde, lave batterinivåindikatoren for batteriladestatus.
3) Kontroller at nye batterier er satt riktig inn i enheten.
4) Hvis de røde og gule advarselslampene blinker, dette betyr at herdelyset har nådd den interne
temperaturbegrensningen. La herdelyset kjøle seg ned i 10 minutter eller bruk et kjølig, fuktig håndkle for å avkjøle
enheten raskt.
5) Hvis den røde advarselslampen blinker og piper kontinuerlig, ring Ultradent kundeservice for reparasjon.

Lyset forblir ikke påslått i ønsket
tidsrom

1) Kontroller modus- og tidtakerlysene for riktig tidsinngang.
2) Kontroller indikatoren for lavt batterinivå for batteriladestatus.
3) Kontroller at nye batterier er satt riktig inn i enheten.

1) Kontroller linsen for gjenværende herdet harpiks/kompositter.
2) Bruk oransje vernebriller med UV-beskyttelse, og kontroller at LED-lysene fungerer.
3) Kontroller effektnivået med lysmåler. Ved bruk av en lysmåler anbefaler Ultradent å kontrollere herdingslyset i standard
effekt-modus.
Lyset herder ikke harpiksen tilstrekkelig
MERK: Den sanne numeriske utgangen vil være skjev på grunn av unøyaktigheten av vanlige lysmålere og den tilpassede LEDpakken som herdingslyset bruker. Lysmålere varierer sterkt, og er designet for spesifikke lysledningstupper og -linser.
4) Sjekk utløpsdatoen på herdende harpiks.
5) Sørg for at riktig teknikk følges i henhold til produsentens anbefalinger.
Batterier vil ikke lade

1) Pass på at batteriene settes i laderen i riktig retning og la batteriene lade opp i 1-3 timer.
2) Hvis røde lys på laderen ikke skifter til grønt, ring Ultralent kundeservice for å bestille erstatningsbatterier og/eller lader.
3) Hvis det ikke er synlige grønne eller røde lys på laderen, kan du ringe Ultradent kundeservice for å bestille eller byttelader og/eller strømadapter.

Laderen lader ikke batteriene

1) Pass på at laderen er koblet til og at strømadapteren er koblet til et strømuttak.
2) Hvis grønne eller røde lys på laderen ikke er synlige, ring Ultradent kundeservice for ny lader og/eller strømadapter.

Kan ikke endre modus eller
tidsintervaller

1) Hold både tids-/modus- og strømknappene nede til en serie med pip indikerer at herdingsyset er låst opp.

Herdingslyset er beregnet for bruk i det elektromagnetiske miljøet som er angitt nedenfor. Kunden eller brukeren må sørge for at den brukes i et slikt miljø.
IMMUNITETSTEST

IEC 60601 testnivå

Samsvarsnivå

Elektromagnetisk miljøveiledning

Elektrostatisk utladning (ESD)± 8 kV kontakt± 15 kV luft

± 8 kV kontakt± 15 kV luft

Fysisk miljø bør begrenses til følgende:
1) IP-kode: IP20
2) Ikke senk i væske.
3) Ikke bruk i nærheten av brannfarlig gass.
Enheten er ikke-APG og ikke-AP.
4) Fuktighetsnivå for oppbevaring: 10 % – 95 %5) Temperaturnivå for oppbevaring: 10 °C – 40 °C

IEC 61000-4-2

Elektrisk rask transient/utløsning
IEC 61000-4-4± 2 kV for strømforsyningslinjer± 2 kV for strømforsyningslinjer± 1 kV for inngangs-/utgangslinjer Merknad 1: Herdingslyset har
ingen I/0-porter

Overspenning
IEC 61000-4-5± 1 kV linje til linje± 2 kV linje til jord

± 1 kV linje til linje± 2 kV linje til jord

Spenning, spenningsfall,kortslutninger, forstyrrelser ogvariasjoner på strømforsyningens
inngangslinjer<5%U(> 95 % fall i U for 0,5 syklus)<5%U(> 95 % fall i U for 0,5 syklus)

40 % U(60 % fall i U for 5 sykluser)

40 % U(60 % fall i U for 5 sykluser)

70 % U(30 % fall i U for 25 sykluser)

70 % U(30 % fall i U for 25 sykluser)

IEC 61000-4-11<5%U(> 95 % fall i U i 5 s)
Strømfrekvens(50/60 Hz) magnetfelt

30 A/m

Herdingslyset er batteridrevet og er ikke i stand tiltilkobling til strømnettet.
Fordi herdingslyset er batteridrevet, er det ikke
gjenstand for elektriske utløsninger, overspenninger,
spenningsfall, kortslutning, avbrudd eller variasjoner
av strømnettet.
Tilbehør-batteriladeren kan være gjenstand for det
ovennevnte, men det er skilt fra og ikke kritisk til
driften av VALO trådløst herdingslys.

Hvis batterispenningen til herdingslyset faller til
4VDC, vil enheten ikke tillate bruk. VALO trådløst
herdingslys slås av. Når nye batterier settes inn og deriktige effektnivåene gjenopprettes, vil herdingslyset
starte på nytt og gå tilbake til samme tilstand før<5%U
strømbruddet. Herdingslyset vil gjenopprettes på(> 95 % fall i U i 5 s)
Merknad 2: Gjenoppretter seg selv egen hånd i tilfelle strømbrudd.
30 A/mIEC 61000-4-8

Effektfrekvensmagnetfelt skal være på nivåer som
er karakteristiske for en typisk plassering i et typisk
bolig-, hjemmepleieomsorgs-, kommersielt, sykehuseller militært miljø.

Merknad 1: Herdingslyset er ikke utstyrt med noen porter eller noen tilgjengelige I/O-linjer.

108

109

Veiledning og produsentens erklæring om elektromagnetisk immunitet for ikke-livsstøttesystemer

Veiledning og produsentens erklæring for anbefalte separasjonsavstander mellom bærbart og mobilt RF kommunikasjonsutstyr og VALO trådløst herdingslys

Herdingslyset er beregnet for bruk i det elektromagnetiske miljøet som er angitt nedenfor. Kunden eller brukeren må sørge for at den brukes i et slikt miljø.

Herdingslyset er beregnet for bruk i et elektromagnetisk miljø der radiostyrt RF-forstyrrelser styres. Brukeren av herdingslyset kan bidra til å forhindreelektromagnetisk interferens ved å opprettholde en minimal avstand mellom bærbart og mobilt RF-kommunikasjonsutstyr (sendere) og herdelyset som anbefalt
nedenfor, i henhold til maksimal utgangseffekt for kommunikasjonsutstyret.

IMMUNITETSTEST

IEC 60601 testnivå

Samsvarsnivå

Ledningsfrekvens

3 Vrms3 VrmsIEC 61000-4-6

150 kHz til 80 MHz

150 kHz til 80 MHz

Elektromagnetisk miljøveiledningBærbart og mobilt RF-kommunikasjonsutstyr bør ikke brukes nærmere noen del av
herdingslyset, inkludert kabler, enn den anbefalte separasjonsavstanden beregnet ut fra
ligningen som gjelder for senderens frekvens.
Anbefalt separasjonsavstand:

80 MHz til 800 MHz

800 MHz til 2,5 GHz
Utstrålt radiofrekvens

3 V/m

3 V/m

P er senderens maksimale utgangseffekt i watt (W) i henhold til senderens produsent, og der anbefalt avstand i meter (m).
Feltstyrker fra faste RF-sendere, som bestemt av en elektromagnetisk stedundersøkelse a,
bør være mindre enn samsvarsnivået i hvert frekvensområde b.

IEC 61000-4-3

80 MHz til 2,5 GHz

80 MHz til 2,5 GHz

Forstyrrelser kan forekomme i nærheten av utstyr merket med følgende symbol:

MERKNAD 1 Ved 80 MHz og 800 MHz vil det høyere frekvensområdet være gjeldende.
MERKNAD 2 Disse retningslinjene gjelder kanskje ikke i alle situasjoner. Elektromagnetisk forplantning påvirkes av absorpsjon og refleksjon fra strukturer,
gjenstander og mennesker.
aFeltstyrker fra faste sendere, for eksempel basestasjoner for radio (mobiltelefoner/trådløse) telefoner og land-mobilradioer, amatørradio, AM- og
FM-radiosendinger og TV-kringkasting kan ikke forutses teoretisk med nøyaktighet. For å vurdere de elektromagnetiske omgivelsene med hensyn til faste
RF-sendere, skal en elektromagnetisk stedsundersøkelse vurderes. Hvis den målte feltstyrken på stedet der herdingslyset er brukt overstiger det gjeldende RFoverensstemmelsesnivået over, bør VALO-ledningsfritt herdingslys observeres for å verifisere normal drift. Hvis unormal ytelse observeres, kan det være nødvendig
med ytterligere tiltak, for eksempel å omorientere eller flytte herdingslyset.
bOver frekvensområdet 150 kHz til 80 MHz bør feltstyrker være mindre enn 3 V/m.

110

Avstand i henhold til senderens frekvens(meter)
Klassifisert maksimal utgangseffekt
av sender(P i watt)

150 kHz – 80 MHz

80 MHz – 800 MHz

800 MHz – 2,5 GHz

0,01

0,12 meter

0,035 meter

0,07 meter

0,1

0,37 meter

0,11 meter

0,22 meter

1

1,7 meter

0,35 meter

0,7 meter

10

3,7 meter

1,11 meter

2,22 meter

100

11,7 meter

3,5 meter

7,0 meter

Herdingslyset er testet i henhold til IEC 60601-1-2:2014 og bestått under utstrålede feltstyrker på 10 V/m mellom 80 MHz til 2,5 GHz. Verdien på 3 Vrms tilsvarer V1
og verdien 10 V/m tilsvarer E1 i formlene ovenfor.
For sendere som er klassifisert med en maksimal utgangseffekt som ikke er nevnt ovenfor, kan den anbefalte avstanden d i meter (m) estimeres ved å bruke
ligningen som gjelder for senderens frekvens, hvor P er maksimal utgangseffekt for senderen i watt (W), i henhold til senderens produsent.
MERK 1 Ved 80 MHz og 800 MHz gjelder separasjonsavstanden for det høyere frekvensområdet.
MERKNAD 2 Disse retningslinjene gjelder kanskje ikke i alle situasjoner. Elektromagnetisk forplantning påvirkes av absorpsjon og refleksjon fra strukturer,
gjenstander og mennesker.

111

VALO™ GRAND

ELΛΥΧΝΙΑ ΠΟΛΥΜΕΡΙΣΜΟΥΠεριγραφή προϊόντος

Με το ευρυζωνικό του φάσμα, το φως πολυμερισμού VALO Grand έχει σχεδιαστεί για να πολυμερίζει όλα τα προϊόντα φωτοπολυμερισμού στην περιοχή μήκους κύματος 385-515 nm ανά
ISO 10650. Το φως πολυμερισμού VALO Grand χρησιμοποιεί τις επαναφορτιζόμενες μπαταρίες Ultradent VALO και τον φορτιστή μπαταριών. Το φως πολυμερισμού έχει σχεδιαστεί για ναστηρίζεται σε μια τυπική βάση οδοντιατρικής μονάδας ή μπορεί να τοποθετηθεί κατά παραγγελία χρησιμοποιώντας τη βάση που περιλαμβάνεται στο κιτ.

Μέρη της μονάδας VALO Grand:
•
•
•
•
•
•

1 – Φως πολυμερισμού VALO Grand4 – Επαναφορτιζόμενες μπαταρίες Ultradent VALO
1 – Φορτιστής μπαταρίας Ultradent VALO με τροφοδοτικό AC 12VDC ιατρικής ποιότητας
1 – Πακέτο δειγμάτων VALO Μανίκι φραγμού
1 – VALO Μεγάλη προστατευτικό ακτινοβολίας
1 – Στήριγμα στερέωσης ελαφριάς επιφάνειας με διπλή κολλητική ταινία

Επισκόπηση των στοιχείων ελέγχου::Φωτογραφία

Οδηγίες βήμα-βήμαΠροετοιμασία

1) Φορτίστε τις μπαταρίες πριν χρησιμοποιήσετε το φως πολυμερισμού (Βλ. ενότητα Συντήρηση μπαταρίας).
2) Πριν από κάθε χρήση, τοποθετήστε ένα νέο προστατευτικό κάλυμμα πάνω από την συσκευή φωτοπολυμερισμού, ελαχιστοποιώντας τις τσακίσεις πάνω από το φακό για καλύτερα
αποτελέσματα.

Σημείωση:Κουμπιά και Ενδεικτικές λυχνίες

LegendΠλήκτρο(α) τροφοδοσίας
(Πάνω ή Κάτω)Φώτα χρονισμού
Ένδειξη χαμηλής μπαταρίαςΦως λειτουργίας / κατάστασης
Πλήκτρο αλλαγής χρονισμού/λειτουργίας

(Κάτω)(Επάνω)Για όλα τα προϊόντα που περιγράφονται, πρέπει να διαβάσετε προσεκτικά και να κατανοήσετε όλες τις οδηγίες και τις πληροφορίες στα δελτία δεδομένων ασφάλειας (SDS) πριν από τηχρήση.

Ενδείξεις χρήσης/Σκοπούμενη χρήσηΩς πηγή φωτός για τον πολυμερισμό φωτοενεργοποιούμενων υλικών οδοντικής αποκατάστασης και συγκολλητικών μέσων.

Προειδοποιήσεις και προφυλάξεις
Ομάδα κινδύνου 2ΠΡΟΣΟΧΗ Αυτό το προϊόν εκπέμπει υπεριώδη ακτινοβολία (UV). Μπορεί να προκύψει ερεθισμός των ματιών ή του δέρματος από την έκθεση. Χρησιμοποιήστε κατάλληλα μέσαπροστασίας.
ΠΡΟΣΟΧΗ Αυτό το προϊόν εκπέμπει δυνητικά επικίνδυνη οπτική ακτινοβολία Μην κοιτάζετε απευθείας τη λυχνία κατά τη διάρκεια της λειτουργίας. Μπορεί να έχει επιβλαβή
επίδραση στα μάτια.
• ΜΗΝ κοιτάτε απευθείας στην έξοδο φωτός. Ο ασθενής, ο κλινικός ιατρός και οι βοηθοί θα πρέπει πάντα να φορούν προστασία ματιών από υπεριώδη ακτινοβολία χρώματοςκεχριμπαριού όταν χρησιμοποιείται φως ωρίμανσης.
• Για να αποφευχθεί ο κίνδυνος ηλεκτροπληξίας, δεν επιτρέπεται καμία τροποποίηση αυτού του εξοπλισμού. Χρησιμοποιήστε μόνο το παρεχόμενο τροφοδοτικό Ultradent VALO καιτους προσαρμογείς βύσματος. Εάν αυτά τα εξαρτήματα είναι κατεστραμμένα, μην τα χρησιμοποιήσετε και καλέστε την Εξυπηρέτηση Πελατών της Ultradent για να παραγγείλετε
αντικατάσταση.
• Ο φορητός εξοπλισμός επικοινωνιών ραδιοσυχνοτήτων μπορεί να υποβαθμίσει την απόδοση εάν χρησιμοποιηθεί σε απόσταση μικρότερη από 30 cm (12 ίντσες).
• Χρησιμοποιείτε μόνο εξουσιοδοτημένα αξεσουάρ, καλώδια και τροφοδοτικά για να αποτρέψετε ακατάλληλη λειτουργία, αυξημένες ηλεκτρομαγνητικές εκπομπές ή μειωμένη
ηλεκτρομαγνητική ατρωσία (ανατρέξτε στην ενότητα Ηλεκτρομαγνητικές εκπομπές).
• Για να αποφύγετε τον κίνδυνο ηλεκτρικής πυρκαγιάς που σχετίζεται με το χειρισμό των μπαταριών:⸰ ΜΗΝ ανακατεύετε επαναφορτιζόμενες μπαταρίες με μη επαναφορτιζόμενες μπαταρίες ή άλλους τύπους μπαταριών.
⸰ ΜΗΝ επιχειρήσετε να φορτίσετε μη επαναφορτιζόμενες μπαταρίες. Η φόρτιση μη επαναφορτιζόμενων μπαταριών μπορεί να προκαλέσει τραυματισμό ή υλικές ζημιές.
⸰ ΜΗΝ βάζετε σε αυτόκλειστο ή ψεκάζετε μπαταρίες, επαφές μπαταριών, φορτιστή ή μετασχηματιστή εναλλασσόμενου ρεύματος με υγρό οποιουδήποτε είδους. Εάν εμφανιστεί
διάβρωση στις επαφές του φορτιστή μπαταρίας, καλέστε το τμήμα εξυπηρέτησης πελατών της Ultradent για να παραγγείλετε αντικατάσταση.
⸰ ΜΗΝ φορτίζετε τις μπαταρίες γύρω από εύφλεκτα υλικά.
⸰ ΜΗΝ φυλάσσετε το φορτιστή σε κλινικό ιατρείο.
• Για να αποφύγετε τον κίνδυνο τραυματισμού, ΜΗΝ χρησιμοποιείτε μπαταρίες που είναι διαβρωμένες (σκουριασμένες), με βαθουλώματα, που εκπέμπουν οσμή ή υγρά, έχουν σκιστείή λείπει το περιτύλιγμα ή έχουν υποστεί άλλη ζημιά. Καλέστε την Εξυπηρέτηση Πελατών της Ultradent για να παραγγείλετε μπαταρίες αντικατάστασης.
• Χρησιμοποιείτε μόνο τις συνιστώμενες μπαταρίες. Οι εναλλακτικές μπαταρίες μπορεί να προκαλέσουν δυσλειτουργία.
• Για να αποφύγετε τον κίνδυνο θερμικού ερεθισμού ή τραυματισμού, αποφύγετε τους κύκλους σκλήρυνσης back-to-back και μην εκθέτετε τους μαλακούς ιστούς του στόματος σεκοντινή απόσταση για περισσότερο από 10 δευτερόλεπτα σε οποιαδήποτε λειτουργία. Εάν απαιτούνται μεγαλύτεροι χρόνοι σκλήρυνσης, χρησιμοποιήστε πολλαπλούς κύκλους
σκλήρυνσης με περιόδους ανάπαυσης μεταξύ των κύκλων ή χρησιμοποιήστε ένα προϊόν διπλής ωρίμανσης για να αποφύγετε τη θέρμανση των μαλακών ιστών.
• Να είστε προσεκτικοί κατά τη θεραπεία ασθενών που υποφέρουν από ανεπιθύμητες φωτοβιολογικές αντιδράσεις ή ευαισθησίες, ασθενείς που υποβάλλονται σε χημειοθεραπεία ή112

ασθενείς που λαμβάνουν θεραπεία με φωτοευαισθητοποιητικά φάρμακα.
• Αυτή η μονάδα μπορεί να είναι ευαίσθητη σε ισχυρά μαγνητικά ή στατικά ηλεκτρικά πεδία, τα οποία θα μπορούσαν να διαταράξουν τον προγραμματισμό. Εάν υποψιάζεστε ότι έχεισυμβεί αυτό, αποσυνδέστε τη μονάδα στιγμιαία και, στη συνέχεια, συνδέστε την ξανά στην πρίζα.
• ΜΗΝ σκουπίζετε την συσκευή φωτοπολυμερισμού με καυστικά ή αποτριπτικά καθαριστικά, μην το βάζετε σε αυτόκαυστο ή μην το βυθίζετε σε οποιοδήποτε λουτρό υπερήχων,απολυμαντικό, διάλυμα καθαρισμού ή υγρό. Η μη τήρηση των οδηγιών επεξεργασίας που περιλαμβάνονται μπορεί να καταστήσει την συσκευή φωτοπολυμερισμού μη λειτουργική.
• Για να αποφύγετε την καταστροφή του εξοπλισμού, ΜΗΝ εισάγετε δάχτυλα, όργανα ή άλλα αντικείμενα στη θήκη μπαταριών του φωτιστικού πολυμερισμού.
• Για να αποφύγετε την καταστροφή του εξοπλισμού, ΜΗΝ επιχειρήσετε να καθαρίσετε τις χρυσές επαφές ή οποιοδήποτε μέρος της θήκης της μπαταρίας. Καλέστε την Εξυπηρέτηση
Πελατών της Ultradent εάν υπάρχει ανησυχία.
• Για να αποφευχθεί ο κίνδυνος διασταυρούμενης μόλυνσης, τα προστατευτικά χιτώνια είναι χρήση για έναν ασθενή.
• Για να μειώσετε τον κίνδυνο διάβρωσης, αφαιρέστε το προστατευτικό χιτώνιο μετά τη χρήση.
• Για να μειώσετε τον κίνδυνο υπο-πολυμερισμένων ρητινών, μην χρησιμοποιείτε φως πολυμερισμού εάν ο φακός έχει καταστραφεί.

• Για να αποτρέψετε τη διασταυρούμενη μόλυνση και να διατηρήσετε το οδοντικό σύνθετο υλικό από το να προσκολλάται στην επιφάνεια του φακού και του σώματος της ράβδου,πρέπει να χρησιμοποιείται εγκεκριμένο προστατευτικό περίβλημα Ultradent πάνω από το φως πολυμερισμού VALO με κάθε χρήση. Τα προστατευτικά καλύμματα προορίζονται γιαχρήση από έναν ασθενή.

Ασπίδα φωτοπροστασίας VALO χωρίς καλώδιο:Η ασπίδα προστασίας VALO χωρίς καλώδιο έχει οβάλ σχήμα, μπορεί να περιστραφεί για μέγιστη χρήση και μπορεί να χρησιμοποιηθεί με ένα διαφανές προστατευτικό κάλυμμα.

Χρήση1) Κάθε τρόπος λειτουργίας χρησιμοποιείται για τον πολυμερισμό οδοντιατρικών υλικών με φωτοενεργοποιητές. Δείτε τον Σύντομο οδηγό τρόπων λειτουργίας για τους συνιστώμενους
χρόνους πολυμερισμού.
ΣΗΜΕΙΩΣΗ: Η λυχνία πολυμερισμού είναι προγραμματισμένη για κυκλική μετάβαση από τον τρόπο λειτουργίας Κανονική Ισχύς στον τρόπο λειτουργίας Υψηλή Ισχύς και Επιπλέον Ισχύς
διαδοχικά. Για παράδειγμα, για να μεταβείτε από τον τρόπο λειτουργίας Κανονική Ισχύς στον τρόπο λειτουργίας Επιπλέον Ισχύς, είναι απαραίτητο να περάσετε πρώτα από τον τρόπο
λειτουργίας Υψηλή Ισχύς Plus για να εισέλθετε στον τρόπο λειτουργίας Επιπλέον Ισχύς.
2) Η λυχνία πολυμερισμού αποθηκεύει το χρονικό διάστημα και τον τρόπο λειτουργίας που έχουν χρησιμοποιηθεί πιο πρόσφατα και επανέρχεται σε αυτές τις ρυθμίσεις απόπροεπιλογή κάθε φορά που αλλάζει ο τρόπος λειτουργίας ή αφαιρούνται οι μπαταρίες.

Λειτουργία
ΛΕΙΤΟΥΡΓΙΑ ΠΟΛΥΜΕΡΙΣΜΟΥ: Λειτουργία Κανονικής Ισχύος

ΧΡΟΝΙΚΑ ΔΙΑΣΤΗΜΑΤΑ: 5, 10, 15, και 20 δευτερόλεπτα.
• Πρόκειται για τον προεπιλεγμένο τρόπο λειτουργίας της λυχνίας πολυμερισμού κατά την ΑΡΧΙΚΗ ενεργοποίηση. Η ένδειξη τρόπου λειτουργίας/κατάστασης θα είναι πράσινη και οιτέσσερις πράσινες ενδείξεις χρονομέτρησης θα ανάψουν, υποδεικνύοντας ότι είναι ενεργός ο τρόπος λειτουργίας Κανονική Ισχύς.
• Για να αλλάξετε τα χρονικά διαστήματα, πατήστε γρήγορα το κουμπί Χρόνου/Τρόπου λειτουργίας.
• Πατήστε το κουμπί Παροχής ισχύος για να ενεργοποιηθεί ο πολυμερισμός. Για να σταματήσετε τον πολυμερισμό πριν ολοκληρωθεί ένα χρονικό διάστημα, πατήστε ξανά το κουμπί
Παροχής ισχύος.

ΛΕΙΤΟΥΡΓΙΑ ΠΟΛΥΜΕΡΙΣΜΟΥ: Λειτουργία Υψηλής Ισχύος PlusΧΡΟΝΙΚΑ ΔΙΑΣΤΗΜΑΤΑ: 1, 2, 3, και 4 δευτερόλεπτα.
• Από τον τρόπο λειτουργίας Κανονική Ισχύς, πατήστε το κουμπί Αλλαγής χρόνου/τρόπου λειτουργίας για 2 δευτερόλεπτα και αφήστε το. Η ένδειξη τρόπου λειτουργίας/κατάστασης
θα είναι πορτοκαλί και οι τέσσερις πράσινες ενδείξεις χρονομέτρησης θα αναβοσβήσουν, υποδεικνύοντας ότι είναι ενεργός ο τρόπος λειτουργίας Υψηλή Ισχύς.
• Για να αλλάξετε τα χρονικά διαστήματα, πατήστε γρήγορα το κουμπί Χρόνου/Τρόπου Λειτουργίας.
• Πατήστε οποιοδήποτε από τα πλήκτρα Παροχής ισχύος για εκκίνηση του πολυμερισμού. Για να σταματήσετε τον πολυμερισμό πριν ολοκληρωθεί ένα χρονικό διάστημα, πατήστε
ξανά ένα από τα κουμπιά Παροχής ισχύος.
• Για να επιστρέψετε στον τρόπο λειτουργίας Κανονική Ισχύς, πατήστε το κουμπί Αλλαγής χρόνου/Τρόπου λειτουργίας για 2 δευτερόλεπτα και αφήστε το. Ο κύκλος θα μετακινηθεί
στον τρόπο λειτουργίας Επιπλέον Ισχύς. Πατήστε ξανά το κουμπί για 2 δευτερόλεπτα και αφήστε το. Η ένδειξη τρόπου λειτουργίας/κατάστασης θα είναι πορτοκαλί και οι τέσσεριςπράσινες ενδείξεις χρονομέτρησης θα ανάψουν, υποδεικνύοντας ότι είναι ενεργός ο τρόπος λειτουργίας Κανονική Ισχύς.

ΛΕΙΤΟΥΡΓΙΑ ΠΟΛΥΜΕΡΙΣΜΟΥ: Λειτουργία Επιπλέον Ισχύος

ΧΡΟΝΙΚΟ ΔΙΑΣΤΗΜΑ: Μόνο 3 δευτερόλεπτα (Σημείωση: Ο τρόπος λειτουργίας Επιπλέον Ισχύς έχει καθυστέρηση ασφαλείας 2 δευτερολέπτων στο τέλος κάθε κύκλου πολυμερισμού
για να περιοριστεί η θέρμανση κατά τη διάρκεια διαδοχικού πολυμερισμού. Στο τέλος της καθυστέρησης, εκπέμπεται ηχητικός τόνος που υποδεικνύει ότι η συσκευή είναι έτοιμη γιαπεραιτέρω χρήση).
• Από τη λειτουργία Κανονική Ισχύς, πατήστε το κουμπί Αλλαγής χρόνου/τρόπου λειτουργίας για 2 δευτερόλεπτα, αφήστε το, πατήστε το ξανά για 2 δευτερόλεπτα και αφήστε το. Ηένδειξη τρόπου λειτουργίας/κατάστασης θα αναβοσβήσει με πορτοκαλί χρώμα και οι τρεις από τις ενδείξεις χρονομέτρησης θα αναβοσβήσουν, υποδεικνύοντας ότι είναι ενεργός οτρόπος λειτουργίας Επιπλέον Ισχύς.
• Πατήστε το κουμπί Παροχής ισχύος για να ενεργοποιηθεί ο πολυμερισμός. Για να σταματήσετε τον πολυμερισμό πριν ολοκληρωθεί ένα χρονικό διάστημα, πατήστε ξανά το κουμπί
Παροχής ισχύος.
• Για να επιστρέψετε στον τρόπο λειτουργίας Κανονική Ισχύς, πατήστε το κουμπί Χρόνου/Τρόπου λειτουργίας για 2 δευτερόλεπτα και αφήστε το. Η ένδειξη τρόπου λειτουργίας/κατάστασης θα είναι πράσινη και οι πράσινες ενδείξεις χρονομέτρησης θα ανάψουν, υποδεικνύοντας ότι είναι ενεργός ο τρόπος λειτουργίας Κανονική ισχύς.

Αναστολή λειτουργίας: Η λυχνία πολυμερισμού μεταβαίνει σε Αναστολή Λειτουργίας μετά από 60 δευτερόλεπτα αδράνειας, όπως υποδεικνύεται από αργό αναβόσβημα της ένδειξης
τρόπου λειτουργίας/κατάστασης. Αν σηκώσετε ή ακουμπήσετε τη μονάδα, η λυχνία πολυμερισμού θα επανέλθει αυτόματα στην τελευταία ρύθμιση που χρησιμοποιήσατε. Για ναμεγιστοποιήσετε τη διάρκεια ζωής της μπαταρίας, μη μετακινείτε τη λυχνία πολυμερισμού όταν δεν την χρησιμοποιείτε.

Καθαρισμός

1) Απορρίψτε τα χρησιμοποιημένα προστατευτικά καλύμματα ως συνήθη απόβλητα μετά τη χρήση σε κάθε ασθενή.
2) Βλ. ενότητα «Επεξεργασία».
113Οδηγίες τοποθέτησης βραχίονα στήριξης

1) Ο βραχίονας θα πρέπει να τοποθετηθεί σε επίπεδη επιφάνεια ελεύθερη από λιπαντικές ουσίες.
2) Καθαρίστε την επιφάνεια με αλκοόλη.
3) Ξεκολλήστε το πίσω μέρος της αυτοκόλλητης ταινίας του βραχίονα.
4) Τοποθετήστε τον βραχίονα έτσι ώστε η λυχνία πολυμερισμού να μετακινείται προς τα πάνω κατά την αφαίρεση. Πιέστε με δύναμη για να στερεώσετε τον βραχίονα στη θέση του.

Σύντομος οδηγός τρόπων λειτουργίας:Κανονική Ισχύς
1000 mW/cm2

Τρόπος λειτουργίας

Υψηλή Ισχύς Plus
1600 mW/cm2

Επιπλέον Ισχύς
3200 mW/cm2

Πλήκτρο Παροχής ισχύος
Ενδείξεις LED τρόπου
λειτουργίας/χρονομέτρησης
Κουμπιά Χρόνου
Επιλογές Χρόνου

5s

10s

15s

20s

1s

2s

3s

4s

Μόνο 3sΓια να αλλάξετε τη ρύθμιση
χρόνου

Πατήστε στιγμιαία και αφήστε το κουμπί Χρόνου για κυκλική εναλλαγή στις επιλογές χρόνου.

Για να αλλάξετε τρόπο
λειτουργίας

Πατήστε και κρατήστε πατημένο το κουμπί Χρόνου για 2 δευτερόλεπτα και αφήστε το. Η λυχνία πολυμερισμού θα μεταβεί
κυκλικά στον επόμενο τρόπο λειτουργίας.

ΥπόμνημαΣταθερά αναμμένες ενδείξεις LEDΠαλλόμενες ενδείξεις LEDΛειτουργία
φωτοπολυμερισμού

Ισχύς * (mW)Ακτινοβολία * (mW/cm2)

Συνολικός χρόνος έκθεσης
(δευτερόλεπτα)Ενέργεια** (Joules)Πρότυπο970

900

5/10/15/20

4,8J/9,7J/14,5J/19,4J

Σύντομος οδηγός πολυμερισμού:High Power Plus

1,620

1,500

1/2/3/4

1,6J/3,2J/4,8J/6,5J

Xtra

2,260

2,100

3

6,8J

*Ονομαστική έξοδος σε απόσταση 2 mm από την άκρη του φακού μέχρι την επάνω επιφάνεια του σύνθετου υλικού.
**Οι τιμές στο τμήμα της συνολικής ενέργειας ανά κύκλο (joules) στρογγυλοποιούνται στο πλησιέστερο δέκατο.
Σύντομος οδηγός προειδοποιήσεων:Προειδοποίηση στάθμης ισχύος
Αντικαταστήστε τις μπαταρίες
• Χαμηλή μπαταρία: αναβοσβήνειαργά• Αυτόματη διακοπή: 3 ηχητικοί
τόνοι, αναβοσβήνει
• Απαγορεύεται η λειτουργία

Προειδοποίηση θερμοκρασίας

Προειδοποίηση επισκευής

Προειδοποίηση LEDΑφήστε τη μονάδα να κρυώσει

Καλέστε την Εξυπηρέτηση Πελατών
για επισκευή

Καλέστε την Εξυπηρέτηση Πελατών
για επισκευή• Χωρίς ηχητικό τόνο• Αναβοσβήνει κάθε 2 δευτερόλεπτα• Επιτρέπεται η λειτουργία

• 3 συνεχόμενοι ηχητικοί τόνοι• Αναβοσβήνει γρήγορα μία φορά• Απαγορεύονται οι διαδικασίες
λειτουργίας

• 3 ηχητικοί τόνοι• Αναβοσβήνει αργά• Απαγορεύεται η λειτουργία

Γενικός καθαρισμός φωτός πολυμερισμού
Μετά από κάθε χρήση, υγράνετε μια γάζα ή ένα μαλακό πανί με ένα εγκεκριμένο απολυμαντικό επιφάνειας και σκουπίστε την επιφάνεια και το φακό. Τα μη εξουσιοδοτημένα
καθαριστικά μπορεί να προκαλέσουν βλάβη στο φως πολυμερισμού.
ΑΠΟΔΕΚΤΑ ΚΑΘΑΡΙΣΤΙΚΑ:• 70% ισοπροπυλική αλκοόλη
• 70% Αιθανόλη
Γενικός καθαρισμός της ασπίδας φωτός:Καθαρίστε τη θωράκιση φωτός VALO Cordless χρησιμοποιώντας οποιοδήποτε απολυμαντικό επιφανειών. ΜΗ βάζετε σε αυτόκαυστο.
Επιδιόρθωση
Συντήρηση που εκτελείται από τον χρήστη
1) Χρησιμοποιήστε ένα προστατευτικό περίβλημα για να αποτρέψετε την προσκόλληση του οδοντικού σύνθετου στην επιφάνεια του φακού. Εάν είναι απαραίτητο, χρησιμοποιήστε έναπλαστικό ή ανοξείδωτο οδοντιατρικό όργανο για να αφαιρέσετε προσεκτικά τυχόν κολλημένο σύνθετο υλικό. Μη χρησιμοποιείτε εργαλεία που θα καταστρέψουν τον φακό.
2) Οι μετρητές φωτός διαφέρουν πολύ και έχουν σχεδιαστεί για συγκεκριμένα ρύγχη οδηγών φωτός και φακούς. Η Ultradent συνιστά τον τακτικό έλεγχο της εξόδου στη λειτουργία
Standard Power. ΣΗΜΕΙΩΣΗ: η πραγματική αριθμητική έξοδος θα παραμορφωθεί λόγω της ανακρίβειας των κοινών φωτόμετρων και του προσαρμοσμένου πακέτου LED στο φωςωρίμανσης
Συντήρηση μπαταρίας
Φόρτιση και αλλαγή μπαταριώνΤο φως πολυμερισμού συνοδεύεται από 4 επαναφορτιζόμενες μπαταρίες φωσφορικού σιδήρου λιθίου. Η Ultradent συνιστά τη χρήση μόνο επαναφορτιζόμενων μπαταριών VALO.
Πώς να φορτίσετε τις μπαταρίες:1) Τοποθετήστε το φορτιστή και τις μπαταρίες σε χώρο όπου δεν θα έρθουν σε επαφή με υγρά ή απολυμαντικό σπρέι.
2) Τοποθετήστε τις μπαταρίες στο φορτιστή με το θετικό (+) άκρο στραμμένο προς τις ενδεικτικές λυχνίες στο φορτιστή.
3) Το πράσινο φως δείχνει ότι οι μπαταρίες είναι έτοιμες για χρήση.
4) Οι μπαταρίες θα χρειαστούν 1-3 ώρες για να φορτιστούν. Αφήστε τις μπαταρίες στο φορτιστή μέχρι να είναι έτοιμες για χρήση.
ΣΗΜΕΙΩΣΗ: Εάν η κόκκινη λυχνία στο φορτιστή δεν ανάβει πράσινη όταν οι μπαταρίες φορτίζονται για περισσότερες από τρεις ώρες, η μπαταρία μπορεί να υποβαθμιστεί (χαλαρώσει)και να μην μπορεί να φορτιστεί. Δοκιμάστε μια νέα μπαταρία ή καλέστε την Εξυπηρέτηση Πελατών της Ultradent για να παραγγείλετε ένα νέο σετ επαναφορτιζόμενων μπαταριών.
Πώς να αλλάξετε/τοποθετήσετε μπαταρίες:1) Αφαιρέστε το πίσω καπάκι στρίβοντας αριστερόστροφα ένα τέταρτο της στροφής.
2) Αφαιρέστε τις μπαταρίες.
3) Τοποθετήστε πρώτα τις νέες μπαταρίες με τη θετική (+) πλευρά.
4) Τοποθετήστε ξανά το πίσω καπάκι ευθυγραμμίζοντας και πιέζοντας απαλά ενώ στρίβετε δεξιόστροφα. Το καπάκι θα κάνει κλικ όταν συνδεθεί πλήρως.
5) Η μονάδα είναι έτοιμη για χρήση.
Χαμηλές μπαταρίες: Η λυχνία ωρίμανσης σηματοδοτεί τον χρήστη ότι είναι ώρα να αλλάξει τις μπαταρίες όταν η ενδεικτική λυχνία χαμηλής μπαταρίας αναβοσβήνει με κόκκινο χρώμα.
Εάν η φόρτιση της μπαταρίας γίνει πολύ χαμηλή, θα ακουστεί ένας ηχητικός προειδοποιητικός ήχος 3 μπιπ και η λυχνία ωρίμανσης δεν θα επιτρέψει περαιτέρω λειτουργία μέχρι ναεπαναφορτιστούν οι μπαταρίες ή να τοποθετηθούν νέες μπαταρίες. (Δείτε τον Οδηγό Γρήγορης Προειδοποίησης)Χρόνος φόρτισης και διάρκεια ζωής μπαταρίας: Η διάρκεια ζωής της πλήρως φορτισμένης μπαταρίας στο φως ωρίμανσης εξαρτάται από το διάστημα λειτουργίας/χρόνου, τον τύπο τηςμπαταρίας, την ποσότητα χρήσης και την απόδοση LED.
• Συνιστώμενο διάστημα επαναφόρτισης: Επαναφορτίστε εάν χρειάζεται ή όταν ανάψει η ένδειξη χαμηλής μπαταρίας. Το προσδόκιμο ζωής εξαρτάται από τη χρήση.
• Επιπλέον μπαταρίες: Το φως πολυμερισμού συνοδεύεται από 4 επαναφορτιζόμενες μπαταρίες.
Επιδιόρθωση από τον κατασκευαστή
1) Οι επισκευές πρέπει να εκτελούνται μόνο από εξουσιοδοτημένο προσωπικό σέρβις. Το Ultradent είναι να παρέχει στο προσωπικό σέρβις τεκμηρίωση για την εκτέλεση επισκευών.
2) Όταν στέλνετε μονάδες για επισκευή, σέρβις ή βαθμονόμηση, αφαιρείτε πάντα τις μπαταρίες από το φως πολυμερισμού και το φορτιστή. Τυλίξτε τις μπαταρίες, τον φορτιστή, τονπροσαρμογέα και το φως ωρίμανσης ξεχωριστά στο κουτί επιστροφής.
3) Αποστολή μπαταριών σύμφωνα με τους τοπικούς κανονισμούς.
ΕγγύησηΗ Ultradent Products, Inc. ("Ultradent") εγγυάται ότι αυτό το προϊόν, για περίοδο 5 ετών από την ημερομηνία αγοράς, όταν λειτουργεί σύμφωνα με τις οδηγίες λειτουργίας πουπεριλαμβάνονται με το προϊόν, (i) θα συμμορφώνεται από κάθε υλική άποψη με την τις προδιαγραφές που ορίζονται στην τεκμηρίωση της Ultradent που συνοδεύει το προϊόν· και (ii) ναμην έχει ελαττώματα στο υλικό και την κατασκευή.
Αυτή η περιορισμένη εγγύηση δεν μεταβιβάζεται και ισχύει αποκλειστικά για τον αρχικό αγοραστή και δεν επεκτείνεται στους επόμενους κατόχους του προϊόντος. Αυτή η περιορισμένη
εγγύηση δεν καλύπτει άλλα αξεσουάρ, όπως, ενδεικτικά, μπαταρίες, φορτιστές, προσαρμογείς ή προσαρμοστικούς φακούς. Αυτή η περιορισμένη εγγύηση είναι άκυρη εάν το προϊόν
αποτύχει ή καταστραφεί λόγω αμέλειας, κατάχρησης, κακής χρήσης, ατυχήματος, τροποποίησης, παραβίασης, αλλοίωσης ή μη τήρησης των ισχυουσών οδηγιών χρήσης. Μόνο γιαλόγους παραδείγματος, ένα προϊόν που έχει πέσει και έχει καταστραφεί δεν καλύπτεται από αυτήν την εγγύηση. Για να πληροίτε τις προϋποθέσεις για αυτήν την περιορισμένη εγγύηση,πρέπει να υποβληθεί στην Ultradent η απόδειξη αγοράς (π.χ. απόδειξη πώλησης ή παρόμοια τεκμηρίωση) μαζί με το ελαττωματικό προϊόν.
Ένα ελαττωματικό προϊόν που πληροί τις προϋποθέσεις εγγύησης που ορίζονται στο παρόν, κατά την αποκλειστική κρίση της Ultradent, είτε θα επισκευαστεί είτε θα αντικατασταθεί. Σεκαμία περίπτωση η ευθύνη της Ultradent για το προϊόν δεν θα υπερβαίνει την τιμή αγοράς που καταβάλλει ο αγοραστής. Σε καμία περίπτωση η Ultradent δεν φέρει ευθύνη για τυχόν
έμμεσες, τυχαίες, προβλεπόμενες, απρόβλεπτες, ειδικές ή αποθετικές ζημιές που προκύπτουν από ή σε σχέση με τη χρήση αυτού του προϊόντος.

Επεξεργασία

Αυτό το προϊόν προορίζεται για χρήση με προστατευτικό χιτώνιο. Δείτε παραπάνω για συντήρηση και γενικό καθαρισμό.

Αποθήκευση και Απόρριψη
Συντήρηση
114

Σε περίπτωση αποθήκευσης της λυχνίας πολυμερισμού για περιόδους που υπερβαίνουν τις 2 εβδομάδες ή τοποθέτησής της σε συσκευασία μεταφοράς, θα πρέπει να αφαιρείτε πάντα
τις μπαταρίες. Εάν οι μπαταρίες παραμείνουν στη μονάδα για μεγάλο χρονικό διάστημα χωρίς επαναφόρτιση, ενδέχεται να μην λειτουργούν ή να μην φορτίζονται. Μην αποθηκεύετε τιςμπαταρίες σε θερμοκρασίες πάνω από 60°C (140°F) ή άμεσα εκτεθειμένες στο ηλιακό φως.
115Θερμοκρασία αποθήκευσης και μεταφοράς της λυχνίας πολυμερισμού:• Θερμοκρασία: +10°C έως +40°C (+50°F έως +104°F)• Σχετική υγρασία: 10% έως 95%• Πίεση περιβάλλοντος: 500 hPa έως 1.060 hPaΧαρακτηριστικό

Διάγραμμα σύγκρισης ονομαστικής εκπομπής ακτινοβολίας

Κατά την απόρριψη ηλεκτρονικών αποβλήτων (λυχνίες πολυμερισμού, φορτιστές, μπαταρίες και τροφοδοτικά), ακολουθήστε τις τοπικές οδηγίες για τα απόβλητα και την ανακύκλωση.

Όργανο μέτρησης

Τεχνικά ΘέματαvΒοηθητικά εξαρτήματαΕίδοςΠληροφορίες CEΠίνακας έντασης
φωτόςΑσπίδα φωτοπροστασίας VALO
χωρίς καλώδιο

Φορτιστής VALO

Μπαταρίες VALO

Τεχνικές πληροφορίες/Δεδομένα

Αποτελεσματικές ζώνες μήκους κύματος πολυμερισμό σύνθετου υλικού:ΦΑΣΜΑΤΙΚΗ ΑΚΤΙΝΟΒΟΛΙΑ (mW/nm)

Αποτελεσματική ζώνηανώτερου μήκους
κύματος
440-480 nmΚΥΜΑ ENGTH (nm)Xtra‡ Αναλυτής φάσματος GigahertzΕκπομπή

Ολική ισχύς

Διάφραγμα μετρητή

15 mm

15 mm

Κανονική Ισχύς
(±10%)

900 mW/cm²

970 mWΥψηλή Ισχύς Plus(±10%)

1500 mW/cm²

1615 mWΕπιπλέον Ισχύς
(±10%)

2100 mW/cm²

2260 mWΤο Radiant Exitance θα ποικίλλει ανάλογα
με την ικανότητα του οργάνου, τη μέθοδο
μέτρησης και την τοποθέτηση φωτός.
‡ Το Radiant Exitance συμμορφώνεται μετο ISO 10650 όταν μετράται με αναλυτή
φάσματος Gigahertz-Optik.

Λυχνία πολυμερισμού VALO GrandΒάρος:
Ονομαστικές τιμές: IEC 60601-1 (Ασφάλεια), IEC 60601-1-2• Με μπαταρίες: 6 ουγγιές (190 γραμμάρια)(ΗΜΣ)• Χωρίς μπαταρίες: 5 ουγγιές (150 γραμμάρια)Διαστάσεις: (8 x 1,28 x 1,06) ίντσες, (203 x 32,5 x 27) mm

Τροφοδοτι-κόφορτιστή

Έξοδος – 12 VDC στα 500 mAΕίσοδος – 100 VAC έως 240 VACΚωδικός προϊόντος (P/N) Ultradent: 5930 Τροφοδοτικό
φορτιστή VALO με βύσματα για διεθνή χρήση

Φορτιστής VALO

Έξυπνος φορτιστής μπαταριών λιθίου φωσφορικού σιδήρου VALO 3,6 VDC:
• Αυτόματη διακοπή κατά την πλήρη φόρτιση• Αυτόματη ανίχνευση ελαττωματικών μπαταριών• Προστασία: Θερμική, Υπερφόρτισης, Βραχυκυκλώματος, αντίστροφης πολικότητας⸰ Κόκκινη ένδειξη LED – Εκτελείται φόρτιση⸰ Πράσινη ένδειξη LED – Άδεια ή πλήρως φορτισμένη
⸰ Σβηστή LED – βραχυκύκλωμα
• Χρόνος φόρτισης: 1 – 3 ώρεςΟνομαστικές τιμές: CE, WEEE

Μπαταρίες VALO

Επαναφορτιζόμενες: Χημικά ασφαλές λίθιο φωσφορικού σιδήρου (LiFePO4) RCR123A

Συνθήκες
λειτουργίας

Θερμοκρασία: +10°C έως +32°C (+50°F έως +90°F)Σχετική υγρασία: 10% έως 95%Ατμοσφαιρική πίεση: 700 hPa έως 1.060 hPaΚύκλος λειτουργίαςΗ λυχνία πολυμερισμού έχει σχεδιαστεί για βραχυπρόθεσμη λειτουργία. Στη μέγιστη θερμοκρασία περιβάλλοντος (32°C) 1 λεπτό ενεργής
λειτουργίας (ON) σε αλλεπάλληλους κύκλους, 30 λεπτά απενεργοποίησης (OFF) (περίοδος ψύξης).

ΑΠΟΤΕΛΕΣΜΑΤΙΚΕΣ ΣΥΝΘΕΤΕΣ ΖΩΝΕΣ ΜΗΚΟΥΣ ΚΥΜΑΤΟΣ ΠΛΗΡΩΣΗΣ
(Εύρος μηκών κύματος που μπορεί να χρησιμοποιηθεί: 385-515 nm)Αποτελεσματικό
χαμηλότερο
ζώνη μήκους κύματος
395-415 nm

Πληροφορίες/Προδιαγραφές

Υψηλή ισχύς
ΠρότυποΟνομαστικές τιμές: IEC 60601-1 (Ασφάλεια)Μήκος καλωδίου - 6 πόδια (1,8 μέτρα)Το τροφοδοτικό του φορτιστή VALO είναι τροφοδοτικό ιατρικής κατηγορίας ΙΙ καιπαρέχει απομόνωση από το ΗΛΕΚΤΡΙΚΟ ΔΙΚΤΥΟΑντιμετώπιση προβλημάτωνΕάν οι λύσεις που προτείνονται παρακάτω δεν διορθώνουν το πρόβλημα, καλέστε την Ultradent στο 800.552.5512. Εκτός των Ηνωμένων Πολιτειών, καλέστε τονδιανομέα ή τον εμπορικό αντιπρόσωπο οδοντιατρικών ειδών Ultradent.

Χαρακτηριστικό

Πληροφορίες/Προδιαγραφές

ΦακόςΔιάμετρος 11,7 mm

Εύρος μήκους
κύματος

116• Ωφέλιμο εύρος μήκους κύματος: 385 – 515 nm• Μέγιστο μήκος κύματος: 395 – 415 nm και 440 – 480 nmΠρόβλημαΠιθανές λύσειςΗ λυχνία δεν ενεργοποιείται

1) Πιέστε το πλήκτρο Χρόνου/Τρόπου Λειτουργίας ή Παροχής Ισχύος για να επαναφέρετε τη λυχνία από τη λειτουργία
εξοικονόμησης ενέργειας.
2) Ελέγξτε την κόκκινη ένδειξη χαμηλής μπαταρίας για την κατάσταση φόρτισης της μπαταρίας.
3) Βεβαιωθείτε ότι έχουν εισαχθεί σωστά νέες μπαταρίες στη μονάδα.
4) Εάν αναβοσβήνουν με κόκκινο και κίτρινο χρώμα οι προειδοποιητικές ενδείξεις LED , αυτό σημαίνει ότι λυχνία
πολυμερισμού έχει φτάσει στο εσωτερικό όριο ασφαλούς θερμοκρασίας. Αφήστε τη λυχνία πολυμερισμού να κρυώσει
για 10 λεπτά ή χρησιμοποιήστε μια δροσερή, υγρή πετσέτα για γρήγορη ψύξη της μονάδας.
5) Εάν η κόκκινη προειδοποιητική ένδειξη LED αναβοσβήνει και εκπέμπει συνεχείς ηχητικούς τόνους, καλέστε τηνΕξυπηρέτηση Πελατών Ultradent για επισκευή.

Η λυχνία δεν παραμένει ενεργή για τηνεπιθυμητή ώρα1) Ελέγξτε τις ενδείξεις τρόπου λειτουργίας και χρονομέτρησης και βεβαιωθείτε ότι έχει επιλεγεί το σωστό χρονικό
διάστημα.
2) Ελέγξτε την ένδειξη χαμηλής μπαταρίας για την κατάσταση φόρτισης της μπαταρίας.
3) Βεβαιωθείτε ότι έχουν εισαχθεί σωστά νέες μπαταρίες στη μονάδα.
117

Εάν οι λύσεις που προτείνονται παρακάτω δεν διορθώνουν το πρόβλημα, καλέστε την Ultradent στο 800.552.5512. Εκτός των Ηνωμένων Πολιτειών, καλέστε τονδιανομέα ή τον εμπορικό αντιπρόσωπο οδοντιατρικών ειδών Ultradent.
ΠρόβλημαΠιθανές λύσειςΗ λυχνία δεν πολυμερίζει σωστάτις ρητίνες

1) Ελέγξτε τον φακό για υπολείμματα ρητίνης/σύνθετων υλικών.
2) Χρησιμοποιώντας τα κατάλληλα πορτοκαλί γυαλιά προστασίας UV, επιβεβαιώστε τη λειτουργία των LED.
3) Ελέγξτε το επίπεδο ισχύος με φωτόμετρο. Εάν χρησιμοποιήσετε φωτόμετρο, η Ultradent συνιστά να ελέγξετε τη λυχνία
πολυμερισμού στον τρόπο λειτουργίας Κανονική Ισχύς.
ΣΗΜΕΙΩΣΗ: Η πραγματική τιμή της ισχύος εξόδου δεν είναι απόλυτα ακριβής λόγω της ανακρίβειας των κοινών φωτόμετρωνκαι της παρουσίας της ειδικής συστοιχίας LED που χρησιμοποιείται στη λυχνία πολυμερισμού. Τα φωτόμετρα διαφέρουν
σημαντικά και είναι σχεδιασμένα για συγκεκριμένα άκρα φωτεινών οδηγών και φακούς.
4) Ελέγξτε την ημερομηνία λήξης της πολυμεριζόμενης ρητίνης.
5) Βεβαιωθείτε ότι ακολουθείται η κατάλληλη τεχνική σύμφωνα με τις συστάσεις του κατασκευαστή.

Οι μπαταρίες δεν φορτίζονται1) Βεβαιωθείτε ότι οι μπαταρίες έχουν τοποθετηθεί στον φορτιστή με τον σωστό προσανατολισμό και αφήστε τις μπαταρίες
να φορτιστούν για 1-3 ώρες.
2) Εάν οι κόκκινες ενδείξεις στον φορτιστή δεν γίνουν πράσινες, καλέστε την Εξυπηρέτηση Πελατών της Ultradent για ναπαραγγείλετε μπαταρίες ή/και φορτιστή αντικατάστασης.
3) Αν δεν είναι ανάβουν ούτε πράσινες ούτε κόκκινες ενδείξεις στον φορτιστή, καλέστε την Εξυπηρέτηση Πελατών Ultradentγια να παραγγείλετε ή να αντικαταστήσετε τον φορτιστή ή/και τον μετασχηματιστή AC.

Ο φορτιστής δεν φορτίζει τις μπαταρίες

1) Βεβαιωθείτε ότι ο φορτιστής είναι συνδεδεμένος στο ρεύμα και ότι ο μετασχηματιστής AC είναι συνδεδεμένος σε μιαπρίζα ρεύματος που λειτουργεί.
2) Αν δεν είναι ανάβουν ούτε πράσινες ούτε κόκκινες ενδείξεις στον φορτιστή, καλέστε την Εξυπηρέτηση Πελατών Ultradentγια νέο φορτιστή ή/και μετασχηματιστή AC.

Δεν είναι δυνατή η αλλαγή τουτρόπου λειτουργίας ή των χρονικών
διαστημάτων

1) Πατήστε ταυτόχρονα τα κουμπιά Χρόνου/Τρόπου λειτουργίας και Παροχής ισχύος έως ότου ακουστεί μια σειρά απόηχητικούς τόνους που υποδεικνύουν ότι η λυχνία πολυμερισμού έχει ξεκλειδωθεί.

Διάφορες πληροφορίες
Οδηγίες και δήλωση κατασκευαστή για ηλεκτρομαγνητικές εκπομπέςΗ λυχνία πολυμερισμού προορίζεται για χρήση στο ηλεκτρομαγνητικό περιβάλλον που καθορίζεται παρακάτω. Ο πελάτης ή ο χρήστης θα πρέπει να εξασφαλίσει ότιχρησιμοποιείται σε τέτοιο περιβάλλον.
Δοκιμή εκπομπών

Συμμόρφωση

Ηλεκτρομαγνητικό περιβάλλον - οδηγίες

Εκπομπές ραδιοσυχνοτήτωνCISPR 11Κατηγορία 1Η λυχνία πολυμερισμού χρησιμοποιεί μπαταρίες και δεν επηρεάζεται από EMI, RF ή καταστολήτης υπέρτασης.

Εκπομπές RF
CISPR 11Κατηγορία BΕκπομπές αρμονικών
IEC 61000-3-2

N/AΗ λυχνία πολυμερισμού χρησιμοποιεί ηλεκτρική και ηλεκτρομαγνητική ενέργεια μόνο για τιςεσωτερικές λειτουργίες. Ως εκ τούτου, οι εκπομπές RF είναι πολύ χαμηλές και είναι απίθανο ναπροκαλέσουν παρεμβολές σε κοντινό ηλεκτρονικό εξοπλισμό.

Διακυμάνσεις τάσης/ασταθείς
εκπομπές IEC 61000-3-3

N/AΗ λυχνία πολυμερισμού είναι κατάλληλη για χρήση σε όλες τις εγκαταστάσεις,
συμπεριλαμβανομένων των οικιακών εγκαταστάσεων και εκείνων που συνδέονται άμεσα με τοδημόσιο δίκτυο παροχής ισχύος χαμηλής τάσης που τροφοδοτεί κτίρια για οικιακή χρήση.

Οδηγίες και δήλωση κατασκευαστή για την ηλεκτρομαγνητική ατρωσίαΗ λυχνία πολυμερισμού προορίζεται για χρήση στο ηλεκτρομαγνητικό περιβάλλον που καθορίζεται παρακάτω. Ο πελάτης ή ο χρήστης θα πρέπει να εξασφαλίσει ότιχρησιμοποιείται σε τέτοιο περιβάλλον.
Δοκιμή ΑΤΡΩΣΙΑΣ

Επίπεδο δοκιμής IEC 60601

Επίπεδο συμμόρφωσης

Οδηγίες για το ηλεκτρομαγνητικό περιβάλλον

Ηλεκτροστατική εκκένωση (ESD)± 8 kV εξ επαφής
± 15 kV από αέρος

± 8 kV εξ επαφής
± 15 kV από αέρος

Ισχύουν οι ακόλουθοι περιορισμοί για το φυσικό
περιβάλλον:1) Κωδικός IP: IP202) Να μην εμβαπτίζεται σε υγρά.
3) Να μη χρησιμοποιείται σε χώρους με εύφλεκτααέρια. Η μονάδα δεν είναι κατάλληλη για χρήση
παρουσία εύφλεκτων αναισθητικών μιγμάτων
(κατηγορίες Non-APG και Non-AP).
4) Εύρος υγρασίας αποθήκευσης: 10% - 95%5) Εύρος θερμοκρασίας αποθήκευσης: 10°C - 40°C

Γρήγορα ηλεκτρικά μεταβατικά
φαινόμενα/απότομες εκφορτίσεις

±2 kV για γραμμές ηλεκτρικής
τροφοδοσίας
± 1 kV για γραμμές εισόδου/εξόδου

±2 kV για γραμμές παροχής
ρεύματος
Σημείωση 1: Η λυχνία
πολυμερισμού δεν έχει θύρες I/0Η λυχνία πολυμερισμού τροφοδοτείται με μπαταρία
και δεν μπορεί να συνδεθεί στο ηλεκτρικό δίκτυο
εναλλασσόμενου ρεύματος.

ΥπέρτασηIEC 61000-4-5± 1 kV γραμμή προς γραμμή
± 2 kV γραμμή προς γείωση

± 1 kV γραμμή προς γραμμή
± 2 kV γραμμή προς γείωση

Τάση, βυθίσεις, βραχυκυκλώματα,διακοπές και διακυμάνσεις στις
γραμμές εισόδου ηλεκτρικής
τροφοδοσίας

<5% U(>95% βύθιση U για 0,5 κύκλους)<5% U(>95% βύθιση U για 0,5 κύκλους)40% U(60% βύθιση U για 5 κύκλους)40% U(60% βύθιση U για 5 κύκλους)70% U(30% βύθιση U για 25 κύκλους)70% U(30% βύθιση U για 25 κύκλους)<5% U(>95% βύθιση U για 5δευτερόλεπτα)<5% U(>95% βύθιση U για 5δευτερόλεπτα)Σημείωση 2: Εκτελεί αυτόματη
επαναφορά

IEC 61000-4-2

IEC 61000-4-11

118Οι δοκιμές εκπομπής αρμονικών και διακυμάνσεων τάσης δεν ισχύουν για τη λυχνία
πολυμερισμού επειδή τροφοδοτείται από μπαταρία.

Επειδή η λυχνία πολυμερισμού τροφοδοτείται μεμπαταρία, δεν υπόκειται σε ηλεκτρικές μεταβάσεις,
υπερτάσεις, βύθιση τάσης, βραχυκυκλώματα,διακοπές ή διακυμάνσεις στην ισχύ του ηλεκτρικού
δικτύου εναλλασσόμενου ρεύματος.
Ο φορτιστής μπαταρίας μπορεί να υπόκειται σταπαραπάνω αλλά είναι ξεχωριστό προϊόν και δεν είναι
κρίσιμος για τη λειτουργία της λυχνίας πολυμερισμού
VALO χωρίς καλώδιο.
Εάν η τάση μπαταρίας της λυχνίας πολυμερισμού
πέσει στα 4 VDC, η μονάδα δεν θα επιτρέψει τηλειτουργία. Η λυχνία πολυμερισμού VALO χωρίς
καλώδιο θα σβήσει. Όταν τοποθετηθούν νέεςμπαταρίες και αποκατασταθούν τα κατάλληλα επίπεδα
ισχύος, η λυχνία πολυμερισμού θα επανεκκινηθεί
και θα επανέλθει στην ίδια κατάσταση πριν από τηναπώλεια ισχύος. Η λυχνία πολυμερισμού εκτελεί
αυτόματη επαναφορά σε περίπτωση απώλειας ισχύος.

119Οδηγίες και δήλωση κατασκευαστή για την ηλεκτρομαγνητική ατρωσία
Συχνότητα ρεύματοςΜαγνητικό πεδίο (50/60 Hz)

30 A/m30 A/mΤα μαγνητικά πεδία συχνότητας ισχύος θα πρέπει
να είναι σε επίπεδα χαρακτηριστικά μιας τυπικής
θέσης σε ένα τυπικό, οικιακό περιβάλλον κατ’ οίκον
φροντίδας ή σε τυπικό εμπορικό, νοσοκομειακό ήστρατιωτικό περιβάλλον.

IEC 61000-4-8Οδηγίες και δήλωση κατασκευαστή για τις συνιστώμενες αποστάσεις διαχωρισμού μεταξύ φορητού και κινητού εξοπλισμού επικοινωνίας RF και της λυχνίας
πολυμερισμού VALO χωρίς καλώδιοΗ λυχνία πολυμερισμού προορίζεται για χρήση σε ηλεκτρομαγνητικό περιβάλλον όπου οι ακτινοβολούμενες διαταραχές RF είναι ελεγχόμενες. Ο χρήστης της λυχνίας
πολυμερισμού μπορεί να βοηθήσει στην πρόληψη των ηλεκτρομαγνητικών παρεμβολών διατηρώντας μια ελάχιστη απόσταση μεταξύ του φορητού και κινητού
εξοπλισμού επικοινωνίας RF (πομποί) και της λυχνίας πολυμερισμού, όπως συνιστάται παρακάτω, ανάλογα με τη μέγιστη ισχύ εξόδου του εξοπλισμού επικοινωνίας.
Απόσταση διαχωρισμού ανάλογα με τη συχνότητα του πομπού
(μέτρα)Σημείωση 1: Η λυχνία πολυμερισμού δεν είναι εξοπλισμένη με θύρες ή με προσβάσιμες γραμμές εισόδου/εξόδου (I/O).

Οδηγίες και δήλωση κατασκευαστή για ηλεκτρομαγνητική ατρωσία σε συστήματα μη υποστήριξης ζωήςΗ λυχνία πολυμερισμού προορίζεται για χρήση στο ηλεκτρομαγνητικό περιβάλλον που καθορίζεται παρακάτω. Ο πελάτης ή ο χρήστης θα πρέπει να εξασφαλίσει ότιχρησιμοποιείται σε τέτοιο περιβάλλον.
Δοκιμή ΑΤΡΩΣΙΑΣ

Αγώγιμες RF

IEC 61000-4-6Επίπεδο δοκιμής IEC
60601

3 Vrms150 kHz έως 80 MHzΕπίπεδοσυμμόρφωσης

3 VrmsIEC 61000-4-3

3 V/m

80 MHz έως 2,5 GHz

80 MHz έως 2,5 GHz

80 MHz – 800 MHz

800 MHz – 2,5 GHz

0,01

0,12 μέτρα

0,035 μέτρα

0,07 μέτρα

0,10,37 μέτρα

0,11 μέτρα

0,22 μέτραΟ φορητός και κινητός εξοπλισμός επικοινωνίας με ραδιοσυχνότητες δεν θα πρέπει
να χρησιμοποιείται πιο κοντά σε οποιοδήποτε μέρος του φωτός σκλήρυνσης,συμπεριλαμβανομένων των καλωδίων, από την συνιστώμενη απόσταση διαχωρισμού πουυπολογίζεται από την εξίσωση που ισχύει για τη συχνότητα του πομπού.
Συνιστώμενη απόσταση διαχωρισμού:1

1,7 μέτρα

0,35 μέτρα

0,7 μέτρα

10

3,7 μέτρα

1,11 μέτρα

2,22 μέτρα

100

11,7 μέτρα

3,5 μέτρα

7,0 μέτρα

150 kHz έως 80 MHz

3 V/m

150 kHz – 80 MHzΟδηγίες για το ηλεκτρομαγνητικό περιβάλλον

80 MHz έως 800 MHzΑκτινοβολούμενες
RFΟνομαστική μέγιστη ισχύς εξόδου
του πομπού
(Ρ σε W)Η λυχνία πολυμερισμού έχει υποβληθεί επιτυχώς σε δοκιμές σύμφωνα με το πρότυπο IEC 60601-1-2:2014 υπό ακτινοβολούμενες εντάσεις πεδίου 10 V/m από 80
MHz έως 2,5 GHz Η τιμή των 3 Vrms αντιστοιχεί στο V1 και η τιμή 10 V/m αντιστοιχεί στο Ε1 στους παραπάνω τύπους.
Για πομπούς με μέγιστη ισχύ εξόδου η οποία δεν αναφέρεται παραπάνω, η συνιστώμενη απόσταση διαχωρισμού d σε μέτρα (m) μπορεί να εκτιμηθεί μέσω τηςεξίσωσης που ισχύει για τη συχνότητα του πομπού, όπου P είναι η μέγιστη ονομαστική ισχύς εξόδου του πομπού σε watt (W) σύμφωνα με τον κατασκευαστή τουπομπού.
ΣΗΜΕΙΩΣΗ 1 Στα 80 MHz και 800 MHz ισχύει η απόσταση διαχωρισμού για το υψηλότερο εύρος συχνοτήτων.
ΣΗΜΕΙΩΣΗ 2 Αυτές οι οδηγίες ενδέχεται να μην ισχύουν σε όλες τις περιπτώσεις. Η ηλεκτρομαγνητική διάδοση επηρεάζεται από την απορρόφηση και τηναντανάκλαση από κατασκευές, αντικείμενα και ανθρώπους.

800 MHz έως 2,5 GHz
P είναι η μέγιστη ονομαστική ισχύς εξόδου του πομπού σε Watt (W) σύμφωνα με τονκατασκευαστή του πομπού και d είναι η συνιστώμενη απόσταση διαχωρισμού σε μέτρα(m).
Οι εντάσεις πεδίου από σταθερούς πομπούς ραδιοσυχνοτήτων, όπως προσδιορίζονται
βάσει επιτόπιας ηλεκτρομαγνητικής μελέτηςα, θα πρέπει να είναι μικρότερες από τοεπίπεδο συμμόρφωσης σε κάθε φάσμα συχνοτήτωνβ.
Είναι πιθανή η δημιουργία παρεμβολών πλησίον εξοπλισμού που φέρει σήμανση με τοακόλουθο σύμβολο:ΣΗΜΕΙΩΣΗ 1 Στα 80 MHz και στα 800 MHz ισχύει το υψηλότερο εύρος συχνοτήτων.
ΣΗΜΕΙΩΣΗ 2 Αυτές οι οδηγίες ενδέχεται να μην ισχύουν σε όλες τις περιπτώσεις. Η ηλεκτρομαγνητική διάδοση επηρεάζεται από την απορρόφηση και τηναντανάκλαση από κατασκευές, αντικείμενα και ανθρώπους.
αΟι τιμές έντασης πεδίου από σταθερούς πομπούς, όπως σταθμοί βάσης ραδιοπομπών (κινητών/ασύρματων), τηλεφώνων και επίγειων φορητών ραδιοπομπών,ερασιτεχνικές ραδιοεκπομπές, ραδιοφωνικές εκπομπές AM και FM και τηλεοπτικές εκπομπές δεν μπορούν να προβλεφθούν θεωρητικά με ακρίβεια. Για τηναξιολόγηση του ηλεκτρομαγνητικού περιβάλλοντος λόγω σταθερών πομπών RF, θα πρέπει να εξεταστεί το ενδεχόμενο διεξαγωγής επιτόπιας ηλεκτρομαγνητικής
μελέτης. Εάν η μετρούμενη ένταση του πεδίου στη θέση όπου χρησιμοποιείται η λυχνία πολυμερισμού υπερβαίνει το παραπάνω ισχύον επίπεδο συμμόρφωσης RF, ημονάδα VALO χωρίς καλώδιο θα πρέπει να παρακολουθείται για να επαληθεύεται η κανονική λειτουργία της. Εάν παρατηρηθεί αφύσικη απόδοση, ίσως χρειαστεί ναληφθούν πρόσθετα μέτρα, όπως η αλλαγή του προσανατολισμού ή της θέσης της λυχνίας πολυμερισμού.
βΓια το εύρος συχνοτήτων από 150 kHz έως 80 MHz, οι τιμές έντασης πεδίου πρέπει να είναι μικρότερες από 3 V/m.

120

121

VALO™ GRAND

BG

ВТВЪРДЯВАЩА СВЕТЛИНА
Описание на продуктаСъс своя широколентов спектър фотополимерната лампа VALO Grand е проектирана за полимеризиране на всички светлинно полимеризируеми продукти в диапазона надължината на вълната от 385-515 nm съгласно ISO 10650. Фотополимерната лампа VALO Grand използва акумулаторни батерии Ultradent VALO и зарядно устройство за батерии.
Фотополимерната лампа е проектирана за стандартна скоба за зъболекарски модул и може да бъде монтирана по поръчка с помощта на скобата, включена в комплекта.

Компоненти за продукта VALO Grand:
•
•
•
•
•
•

Снимка

Бутони и светлини

Бележка:• За да се предотврати кръстосано замърсяване и прилепването на денталния композитен материал към повърхността на лещата и тялото на уреда, за фотополимерната
лампа VALO трябва да се използва одобрена от Ultradent защитна обвивка при всяка употреба. Бариерните пликове са предназначени за употреба от един пациент.

VALO Безжична фото лампа:Безжична фото лампа VALO е с овална форма, може да се върти за максимално използване и може да се използва с прозрачен защитен калъф.

Легенда

Бутон (и) за захранване
(Горната или долната)Осветление на времето
Ниска батерия - индикатор
Режим / състояние на светлината
Бутон за смяна на време
/ режим

(Дъно)(Връх)

За всички описани продукти внимателно прочетете и разберете всички инструкции и SDS информация преди употреба.

Показания за употреба / Предназначение

Източник на осветление за втвърдяване на фотоактивирани стоматологични възстановителни материали и лепила.

Предупреждения и предпазни мерки
Рискова група 2ВНИМАНИЕ! UV лъчи, излъчвани от този продукт. Възпаление на очите или кожата може да е резултат от продължително излагане. Използвайте подходяща лампа.
ВНИМАНИЕ Възможна опасна оптична радиация, излъчвана от този продукт. Не гледайте към работната лампа. Може да бъде вредно за очите.
• НЕ гледайте директно към светлината. Пациентът, клиницистът и асистентите трябва да носят UV защита на очите с кехлибарен цвят при използване на фотополимерната
лампа.
• За да се предотврати рискът от токов удар, не се допуска модифицирането на това оборудване. Използвайте само включеното захранване Ultradent VALO и щепселни
адаптери. Ако тези компоненти са повредени, не ги използвайте и се обадете на отдела за обслужване на клиенти на Ultradent, за да поръчате подмяна.
• Преносимо RF комуникационно оборудване може да влоши производителността, ако се използва на растояние под 30 см (12 инча).
• Използвайте само разрешени аксесоари, кабели, зарядно устройство, батерии и захранващи устройства, за да предотвратите неправилна работа, повишаване наелектромагнитните емисии и намаляване на електромагнитната устойчивост (вижте раздела "Електромагнитни емисии").
• За да избегнете риска от електрически пожар, свързан с батериите:⸰ НЕ съчетавайте акумулаторните батерии с незареждаеми батерии или други видове батерии.
⸰ НЕ се опитвайте да зареждате неакумулаторни батерии. Зареждането на незареждаеми батерии може да доведе до нараняване или материални щети.
⸰ НЕ автоклавирайте и не пръскайте батериите, контактите на батериите, зарядното устройство или захранващия адаптер с какъвто и да е вид течност. Ако се появи
корозия по контактите на зарядното устройство за батерии, обадете се на отдела за обслужване на клиенти на Ultradent, за да поръчате подмяна.
⸰ НЕ зареждайте батериите в близост до запалими материали.
⸰ НЕ дръжте зарядното устройство в операционната.
• За да избегнете риска от нараняване, НЕ използвайте батерии, които са корозирали (ръждясали), имат вдлъбнатини, излъчват миризма или течности, с разкъсана илилипсваща опаковка или са повредени по друг начин. Обадете се на отдела за обслужване на клиенти на Ultradent, за да поръчате нови батерии.
• Използвайте само препоръчани батерии. Други батерии могат да причинят неизправност.
• За да предотвратите риска от термично дразнене или нараняване, избягвайте последователни цикли на полимеризиране и не излагайте оралните меки тъканив непосредствена близост за повече от 10 секунди във всеки режим. Ако са необходими по-дълги периоди на полимеризиране, използвайте множество цикли наполимеризиране с периоди на почивка между циклите или използвайте продукт с двойно полимеризиране, за да избегнете нагряване на меките тъкани.
• Бъдете внимателни, когато лекувате пациенти, които страдат от нежелани фотобиологични реакции или чувствителност, пациенти, които са подложени на химиотерапия, илипациенти, лекувани с фотосенсибилизиращо лекарство.
• Това устройство може да е податливо на силни магнитни или статични електрически полета, които могат да нарушат програмирането му. Ако подозирате, че това се еслучило, изключете уреда от контакта и след това го включете отново.
• НЕ бършете фотополимерната лампа с разяждащи или абразивни почистващи препарати, не автоклавирайте или потапяйте в ултразвукова вана, дезинфектант, почистващ
122

По етапни инструкции
Подготовка

1) Заредете батериите, преди да използвате фотополимерната лампа (вижте раздел "Поддръжка на батерията").
2) Преди всяка употреба поставяйте нов бариерен плик върху фотополимерната лампа, като сведете до минимум гънките върху лещата за най-добри резултати.

1 – Фотополимерна лампа VALO Grand4 – Акумулаторни батерии Ultradent VALO
1 – Зарядно устройство за батерии Ultradent VALO с медицински адаптер за променлив ток 12 VDC
1 – Опаковка за проби от бариерен плик VALO
1 – Светлинен екран VALO Grand1 – Скоба за повърхностен монтаж с двойна залепваща лента

Преглед на контролите:разтвор или течност. Неспазването на инструкциите за обработка може да доведе до неизправност.
• За да избегнете повреда на оборудването, НЕ пъхайте пръсти, инструменти или други предмети в отделението за батерии/клетки на фотополимерната лампа.
• За да избегнете повреда на оборудването, НЕ се опитвайте да почиствате златните контакти или която и да е част от отделението за батерии. Обадете се на отдела заобслужване на клиенти на Ultradent, ако имате въпроси.
• За да се предотврати рискът от кръстосано заразяване, бариерните ръкави не могат да се използват за повече от един пациент.
• За да намалите риска от корозия, отстранете втулка бариерния плик след употреба.
• За да намалите риска от недостатъчно полимеризирани смоли, не използвайте фотополимерната лампа, ако лещите са повредени.

Употреба

1) Всеки режим на захранване се използва за втвърдяване на стоматологичните материали с фото инициатори. Вижте Ръководство за бърз режим за препоръчителните
времена за втвърдяване.
ЗАБЕЛЕЖКА: Лампата за втвърдяване е програмирана да върти последователно от стандартната мощност до високата мощност до режима допълнителна мощност.
Например, за да преминете от режим на стандартна мощност към режим допълнителна мощност, е необходимо да преминете в режим на висока мощност и след това в режим
допълнителна мощност.
2) Лампата за втвърдяване съхранява най-скоро използвания интервал и режим на синхронизация, като по подразбиране ще се върне към това винаги, когато се променят
режимите или ако батериите са отстранени.

Експлоатация
РЕЖИМ НА ВТВЪРДЯВАНЕ: Режим на стандартна мощност

ВРЕМЕННИ ИНТЕРВАЛИ: 5, 10, 15, и 20 секунди.
• Лампата за втвърдяване по подразбиране е в този режим, когато е включена от НАЧАЛО. Индикаторът за режим / състояние ще свети в зелено и четирите зелени светлинина времето ще светят, което показва режим на стандартна мощност.
• За да промените интервалите от време, бързо натиснете бутона Време / Режим.
• Натиснете бутона за захранване, за да се втвърди. За да спрете втвърдяването преди завършването на времевия интервал, натиснете отново бутона за захранване.

РЕЖИМ НА ВТВЪРДЯВАНЕ: Режим с висока мощност плюсВРЕМЕННИ ИНТЕРВАЛИ: 1, 2, 3, и 4 секунди.
• От режим на стандартно захранване натиснете и задръжте бутона за промяна на време / режим за 2 секунди и го пуснете. Индикаторът за режим / състояние ще свети воранжево и четирите зелени светлини на времето ще светят, което показва режим на висока мощност.
• За да промените интервалите от време, бързо натиснете бутона Време / Режим.
• Натиснете бутона за захранване, за да се втвърди. За да спрете втвърдяването преди завършването на времевия интервал, натиснете отново бутона за захранване.
• За да се върнете в режим на стандартно захранване, натиснете и задръжте бутона за промяна на време / режим за 2 секунди и го пуснете, за да преминете към режим
допълнителна мощност. Натиснете и задръжте отново за 2 секунди и пуснете. Индикаторът за режим / състояние ще свети в зелено и четирите зелени светлини на времетоще светят, което показва режим на стандартна мощност.

РЕЖИМ НА ВТВЪРДЯВАНЕ: Режим Допълнителна мощност

ВРЕМЕННИ ИНТЕРВАЛИ: Само 3 секунди (Забележка: Режимът Допълнителна мощност има 2 секундно забавяне на безопасността в края на всеки цикъл на втвърдяванеза ограничаване на затоплянето по време на последователно втвърдяване. В края на закъснението звуковият сигнал показва, че устройството е готово за продължителна
употреба).
• От режим на стандартно захранване натиснете бутона за смяна на времето/режима за 2 секунди, отпуснете, натиснете и задръжте отново за 2 секунди и пуснете.
Индикаторът за режим / състояние ще свети в оранжево и три от зелените светлини на времето ще премигват, което показва режим Допълнителна мощност.
• Натиснете бутона за захранване, за да се втвърди. За да спрете втвърдяването преди завършването на времевия интервал, натиснете отново бутона за захранване.
• За да се върнете в режим на стандартно захранване, натиснете и задръжте бутона за време / режим за 2 секунди и го пуснете. Индикаторът за режим / състояние ще свети взелено и зелените светлини на времето ще светят, което показва режим на стандартна мощност.

Спящ режим: Светлината за втвърдяване ще премине в спящ режим след 60 секунди бездействие, както е посочено от бавното мигане на индикатора за режим / състояние.
Вдигането или докосването на устройството ще събуди светлината за втвърдяване и автоматично ще я върне към последната използвана настройка. За да увеличите живота набатерията, оставете светлината за втвърдяване спокойна, когато не се използва.

Почистване

1) Изхвърлете използваните защитни калъфи в стандартните отпадъци след всеки пациент.
2) Вижте Раздел за Обработка.

Инструкции за монтажни скоби

1) Скобата трябва да се монтира на плоска, безмаслена повърхност.
2) Почистете повърхността със спирт.
3) Отлепете отлепващата лента на скобата.
4) Позиционирайте конзолата така, че светлината за втвърдяване да се вдигне нагоре, когато бъде свалена. Натиснете здраво на място.

Ръководство за бърз режим:123

Стандартна мощност
1000 mW/cm2

РежимРежим с висока мощност плюс1600 mW/cm2

Допълнителна мощност
3200 mW/cm2

Бутон за включване
Светодиоди за режим /синхронизация
Бутони за време
Опции за време

5s

10s

15s

20s

1s

2s

3s

4s

Само 3sЗа да промените времето

Натиснете и отпуснете бутона Time, за да преминете през времеви опции.

За да промените режимите

Натиснете и задръжте бутона за време 2 секунди и го освободете. Светлината за втвърдяване ще премине към следващия
режим.

Легенда

Твърди светодиоди

Мигащи светодиоди

Ръководство за бързо излекуване:Режими наполимеризиране

Мощност * (mW)Излъчване * (mW/cm2)

Общо време на експозиция (секунди)Енергия**(джаули)Standard

970

900

5/10/15/20

4.8J/9.7J/14.5J/19.4J

High Power Plus

1 620

1 500

1/2/3/4

1.6J/3.2J/4.8J/6.5J

Xtra

2 260

2 100

3

6.8J

*Номинална мощност на разстояние 2 мм от върха на лещата до горната повърхност на композита.
**Стойностите в раздела за обща енергия за цикъл (джаули) са закръглени до най-близката десета.
Ръководство за бързо предупреждение:Предупреждение за нивото намощност

Предупреждение за температурата

Сменете батериите

Оставя се да се охлади

• Ниска батерия: бавно мига• Изключи 3 бипкания, мига• Забранява работа

• 3 бипкания
• Бавно мигане
• Забранява работаnПредупреждение за услугата

LED Предупреждение

Свържете се с Обслужване на клиенти Свържете се с Обслужване на клиентиза ремонтза ремонт
• Без звук• Мигане на всеки 2 секунди
• Позволява операция

• Непрекъснати 3 бипкания
• Единична бърза светкавица
• Забранява операции

Ремонт
Поддръжка, извършена от потребителя
1) Използвайте бариерен ръкав, за да предотвратите залепването на дентален композит по повърхността на лещата. Ако е необходимо, използвайте стоматологичен
инструмент от пластмаса или неръждаема стомана, за да отстраните внимателно полепналия композит. Не използвайте инструменти, които могат да повредят лещата.
2) Светломерите се различават значително и са предназначени за специфични световодни накрайници и лещи. Ultradent препоръчва рутинна проверка на изхода врежим Standard Power. ЗАБЕЛЕЖКА: истинският цифров изход ще бъде изкривен поради неточността на обикновените светломери и персонализирания LED пакет въвфотополимерната лампа
Поддръжка на батериите
Зареждане и смяна на батериите
Фотополимерната лампа се предлага с 4 презареждаеми литиево-желязо-фосфатни батерии. Ultradent препоръчва използването само на акумулаторни батерии VALO.
Как се зареждат батериите:1) Поставете зарядното устройство и батериите на място, където няма да влязат в контакт с течности или дезинфектант.
2) Поставете батериите в зарядното устройство с положителния (+) край насочен към светлинните индикатори на зарядното устройство.
3) Зелените светлини показват, че батериите са готови за употреба.
4) Зареждането на батериите отнема 1-3 часа. Оставете батериите в зарядното устройство, докато не станат готови за употреба.
ЗАБЕЛЕЖКА: Ако червената светлина на зарядното устройство не светне в зелено, след като батериите са се зареждали повече от три часа, това може да означава че някоя отбатериите е повредена. Опитайте с нова батерия или се обадете на отдела за обслужване на клиенти на Ultradent, за да поръчате нов комплект акумулаторни батерии.
Как да смените/поставите батериите:1) Отстранете задния капак, като завъртите една четвърт оборот обратно на часовниковата стрелка.
2) Извадете батериите.
3) Първо поставете нови батерии с положителната (+) страна навътре.
4) Поставете отново задния капак, като подравните и леко натиснете, докато завъртате по посока на часовниковата стрелка. Капакът ще щракне, когато е напълно прикрепен.
5) Устройството е готово за употреба.
Изтощени батерии: Фотополимерната лампа сигнализира на потребителя, че е време да смени батериите, когато индикаторът за изтощена батерия мига в червено. Ако зарядътна батерията стане твърде нисък, ще се чуе предупредителен сигнал с 3 звукова и фотополимерната лампа няма да позволи по-нататъшна работа, докато батериите не бъдат
презаредени или не бъдат поставени нови батерии. (Вижте краткото ръководство за предупрежденията
Време за зареждане и живот на батерията: Животът на напълно заредената батерия във фотополимерната лампа зависи от интервала на режим/час (Mode/Time), типа набатерията, честотата на използване и ефективността на светодиода.
• Препоръчителен интервал на презареждане: Презареждайте, когато е необходимо или когато светне индикаторът за изтощена батерия. Продължителността на живота набатерията зависи от начина на употреба.
• Допълнителни батерии: Фотополимерната лампа се доставя с 4 акумулаторни батерии.
Ремонт от производителя
1) Ремонтът трябва да се извършва само от оторизиран сервизен персонал. Ultradent предоставя на сервизния персонал документация за извършване на ремонти.
2) Когато изпращате модули за ремонт, обслужване или калибриране, винаги изваждайте батериите от фотополимерната лампа и зарядното устройство. При изпращане
опаковайте отделно батериите, зарядното устройство, адаптера и фотополимерната лампа.
3) Изпращайте батериите в съответствие с местните разпоредби.
Гаранция
Ultradent Products, Inc. („Ultradent“) гарантира, че този продукт, за период от 5 години от датата на закупуване, ако се използва съгласно инструкциите за експлоатация,включени към продукта, (i) ще отговаря във всички съществени аспекти на спецификациите, посочени в документацията на Ultradent, придружаваща продукта; и (ii) няма даима дефекти в материала и изработката.
Тази ограничена гаранция не подлежи на прехвърляне и се прилага само за първоначалния купувач и не важи за следващите собственици на продукта. Тази ограничена
гаранция не покрива други компоненти на аксесоари като, но не само, батерии, зарядни устройства, адаптери или адаптивни лещи. Тази ограничена гаранция е невалидна, акопродуктът не работи или е повреден поради небрежност, злоупотреба, неправилна употреба, злополука, модификация, подправяне, промяна или неспазване на приложимите
инструкции за употреба. Само за пример, продукт, който е изпуснат и повреден, не се покрива от тази гаранция. За да отговаряте на условията на тази ограничена гаранция, наUltradent трябва да бъде предоставено доказателство за покупка (напр. касова бележка или подобна документация) заедно с дефектния продукт.
Дефектният продукт, отговарящ на гаранционните условия, посочени тук, ще бъде ремонтиран или заменен по усмотрение на Ultradent. В никакъв случай отговорността
на Ultradent за продукта не надвишава покупната цена, платена от купувача. При никакви обстоятелства Ultradent не носи отговорност за косвени, случайни, предвидени,непредвидени, специални или последващи щети, произтичащи от или във връзка с използването на този продукт.

Обработване

Този продукт е предназначен за използване с бариерен ръкав. Вижте по-горе информация за поддръжка и общо почистване.

Съхранение и изхвърляне
Поддръжка

Общо почистване на фотополимерната лампа
След всяка употреба навлажнете марля или мека кърпа с одобрен дезинфектант за повърхности и избършете повърхността и лещата. Неодобрени почистващи препарати могатда причинят повреда на лампата.
ОДОБРЕНИ ПОЧИСТВАЩИ СРЕДСТВА:• 70% изопропилов алкохол
• 70% етанол
Общо почистване на светлинния екран:Почиствайте безжичния светлинен екран VALO с дезинфектант за повърхности. Не автоклавирайте.

124

Ако съхранявате светлината за втвърдяване за период по-дълъг от 2 седмици, или я опаковате за пътуване, винаги изваждайте батериите. Ако батериите са оставени вустройството за дълъг период от време без презареждане, те могат да станат нефункционални или незаредени. Не съхранявайте батериите при температури над 40 ° C (104 ° F)или на пряка слънчева светлина.
Съхранение и транспортиране на светлината за втвърдяване:• Температура: + 10 ° C до + 40 ° C (+ 50 ° F до + 104 ° F)• Относителна влажност: 10% до 95%• Външно налягане: 500 hPa до 1060 hPaПри изхвърляне на електронни отпадъци (напр. Втвърдяващи светлини, зарядни устройства, батерии и захранващи устройства) следвайте местните указания за отпадъци ирециклиране.

Технически съображения
125

Аксесоари

Атрибут
ВещИнформация / Спецификация

Информация за CEНоминална диаграма за сравнение на излъчващото излъчване
Инструмент заизмерване

VALO Безжична фото лампа

Таблица заинтензивност насветлината

VALO зарядно устройство

ExitanceОбща мощност

Бленда на метър

15 mm

15 mm

Стандартна
мощност (±10%)900 mW/cm²

970 mWРежим с висока
мощност плюс(±10%)

1500 mW/cm²

1615 mW

Допълнителна
мощност (±10%)2100 mW/cm²

2260 mW

VALO Батерии

Техническа информация / Данни

Ефективни композитни втвърдяващи се дължини на вълната:СПЕКТРАЛНА МОЩНОСТ НА ЛЪЧЕНИЕ (mW/nm)ЕФЕКТИВНИ ЛЕНТИ НА ДЪЛЖИНА НА ВЪЛНАТА НА ВЪТЪРЖДАВАНЕ НАКОМПОЗИТ
(Използваем диапазон на дължина на вълната: 385-515 nm)Ефективно по-ниско
лента с дължина навълната
395-415 nmАтрибут

Информация / Спецификация

ЛещиДиаметър 11,7 mm

Дължина навълната

‡ Gigahertz спектрален анализатор

Ефективна горна
лента на дължинатана вълната
440-480 nm

WAVEL ENGTH (nm)• Utilizable wavelength range: 385 – 515nm• Peak wavelengths: 395 – 415nm and 440 – 480nm

Xtra
High PowerStandard

Radiant Exitance ще варира в зависимост отвъзможностите на инструмента, метода наизмерване и разположението на светлината.
‡ Radiant Exitance отговаря на ISO
10650, когато се измерва със спектрален
анализатор Gigahertz-Optik.

Световната
лечебна лампа
VALO

Оценки: IEC 60601-1 (Безопасност), IEC 60601-1-2(EMC)Тегло:• С батерии: 6 унции. (190 грама)• Без батерии: 5 унции. (150 грама)Размери: (8 x 1.28 x 1.06) инча (203 x 32.5 x 27) mm

Зарядно
устройство зазахранване

Изход - 12VDC при 500mAВход - 100VAC до 240VAC
Ultradent P / N 5930 Зарядно устройство зазахранване VALO с международни щепселни вложки

Оценки: IEC 60601-1 (Безопасност)Дължина на кабела - 1,8 метра
Захранващото устройство VALO е захранващо устройство за медицински класот клас II и осигурява изолация от захранването MAINS

VALO зарядно
устройство

Зарядно устройство за акумулаторни батерии VALO 3.6VDC Lithium Iron фосфат:• Автоматично изключване при пълно зареждане
• Автоматично откриване на дефектни батерии
• Защити: Термично, презареждане, късо съединение, обратна полярност⸰ Червен светодиод - зареждане⸰ Зелен светодиод - празен или напълно зареден
⸰ LED изключен - късо съединение
• Време за зареждане: 1 - 3 часаРейтинг: CE, WEEEVALO БатерииС акумулаторна батерия: Безопасна химия Литиев железен фосфат (LiFePO4) RCR123A• Работно напрежение: 3.2VDC
Оценки: UL, CE, RoHS, WEEEУсловия за работа

Температура: + 10 ° C до + 32 ° C (+ 50 ° F до + 90 ° F)Относителна влажност: 10% до 95%Външно налягане: 700 hPa до 1060 hPaДежурен цикъл

Светлината за втвърдяване е предназначена за краткотрайна работа. При максимална температура на околната среда (32 ° C) 1
минута ON на обратна връзка, 30 минути OFF (период на охлаждане).

Отстраняване на проблемиАко предложените по-долу решения не отстранят проблема, моля, обадете се на Ultradent на тел. 800.552.5512. Извън Съединените щати се обадете надистрибутора на Ultradent или на стоматологичен дилър.
Проблем

Светлината няма да се включи

126Възможни решения
1) Натиснете бутона за промяна на време / режим или бутон за захранване, за да го събудите от режим на спестяванена енергия.
2) Проверете червения индикатор за батерията за състоянието на зареждане на батерията.
3) Проверете дали новите батерии са поставени правилно в устройството.
4) Ако мигат червени и жълти предупредителни светодиоди, това означава, че светлината за втвърдяване едостигнала границата на безопасност за вътрешна температура. Оставете светлината за втвърдяване да се охлади
за 10 минути или използвайте хладка влажна кърпа, за да охладите устройството бързо.
5) Ако червеният предупредителен светодиод премигва непрекъснато, обадете се на отдела на Ultradent заОбслужване на клиенти за ремонт.

127

Ако предложените по-долу решения не отстранят проблема, моля, обадете се на Ultradent на тел. 800.552.5512. Извън Съединените щати се обадете надистрибутора на Ultradent или на стоматологичен дилър.
Проблем

Възможни решения

Разнородна информация
Ръководство и Декларация за производство на електромагнитни емисии
Светлината за втвърдяване е предназначена за използване в електромагнитната среда, посочена по-долу. Клиентът или потребителят трябва да се уверят, четой се използва в такава среда.

Светлината не остава включена зажеланото време

1) Проверете режима и таймерите за правилно въвеждане на време.
2) Проверете индикатора за батерията за състоянието на зареждане на батерията.
3) Проверете дали новите батерии са поставени правилно в устройството.

Тест за емисии

съгласие

Електромагнитна среда - насоки

RF емисииCISPR 11Група 1Светлината за втвърдяване използва батерии и не е засегната от EMI, RF или пренапрежение.

Светлината не втвърдява смолата
правилно

1) Проверете обектива за остатъчни смоли / композити.
2) Използвайки подходяща кехлибарена UV защита за очите, проверете дали LED светлините работят.
3) Проверете нивото на мощност със светломер. Ако използвате измервателен уред за светлина, Ultradentпрепоръчва да проверявате светлината за втвърдяване в режим на стандартно захранване.
ЗАБЕЛЕЖКА: Истинската цифрова продукция ще бъде изкривена поради неточността на обикновените светлинни
измервателни уреди и обичайния LED пакет, използван от VALO. Измервателите на светлина се различават значителнои са предназначени за специфични светлинни накрайници и лещи.
4) Проверете срока на годност върху втвърдителната смола.
5) Уверете се, че се спазва правилната техника съгласно препоръките на производителя.

RF емисияCISPR 11Клас Б

Втвърдяващата светлина използва електрическа и електромагнитна енергия само за своите
вътрешни функции. Поради това всички радиочестотни емисии са много ниски и нямавероятност да предизвикат смущения в близкото електронно оборудване.

Хармонични емисии
IEC 61000-3-2

N/AТестването за хармонично излъчване и флуктуация на напрежение не е приложимо засветлината на втвърдяване, защото е захранвана с батерии.

Флуктуации на напрежение
/ емисии на трептене IEC
61000-3-3

N/AБатериите няма да се зареждат

1) Уверете се, че батериите са поставени в зарядното устройство в правилната ориентация и позволяват батериитеда се зареждат за 1-3 часа.
2) Ако червените светлини на зарядното устройство не се променят на зелено, обадете се на Службата за обслужванена клиенти на Ultradent, за да поръчате резервни батерии и / или зарядно устройство.
3) Ако не се виждат зелени или червени светлини на зарядното устройство, обадете се на Обслужване на клиенти на
Ultradent, за да поръчате или замените зарядното устройство и / или променливотоковия адаптер.

Светлината за втвърдяване е подходяща за използване във всички обекти, включително
домашни предприятия и тези, които са директно свързани с обществената нисковолтова
електрозахранваща мрежа, която доставя сгради за домашна употреба.

Зарядното устройство не зарежда
батериите

1) Уверете се, че зарядното устройство е включено и променливотоковият адаптер е включен в работещ контакт.
2) Ако не се виждат зелени или червени светлини на зарядното устройство, обадете се за обслужване на клиенти на
Ultradent за ново зарядно устройство и / или променливотоков адаптер.

Светлината за втвърдяване е предназначена за използване в електромагнитната среда, посочена по-долу. Клиентът или потребителят трябва да се уверят, четой се използва в такава среда.

Не може да се променя режимът илиинтервалите от време

1) Задръжте бутоните за време / режим и захранването надолу, докато серия от сигнали показва, че светлината завтвърдяване не е отключена.

Декларация за електромагнитния имунитет

Тест IMMUNITY

IEC 60601 ниво на изпитване

Ниво на съответствие

Ръководство за електромагнитна среда

Електростатичен разряд (ESD)± 8 kV контакт
± 15 kV въздух

± 8 kV контакт
± 15 kV въздух

Физическата среда следва да се ограничи доследното:1) IIP код: IP202) Не потапяйте в течност.
3) Не използвайте около запалим газ. Единицата
е Non-APG и Non-AP.
4) Диапазон на влажност на съхранение:10% - 95%5) Диапазон на температурите на съхранение:10 ° С - 40 ° С

IEC 61000-4-2Електрически бързо преходно
/ избухване
IEC 61000-4-4± 2 kV за захранващи линии
± 2 kV за захранващи линии
± 1 kV за входно / изходни линии Бележка 1: Лампата завтвърдяване няма I / 0 портове

вълнение
IEC 61000-4-5± 1 kV линия към линия
± 2 kV линия към земята

± 1 kV линия към линия
± 2 kV линия към земята

Електрически бързо преходно
/ избухване
IEC 61000-4-4<5% U(>95% dip in U for 0.5 cycle)<5% U(>95% dip in U for 0.5 cycle)

40% U(60% dip in U for 5 cycles)

40% U(60% dip in U for 5 cycles)

70% U(30% dip in U for 25 cycles)

70% U(30% dip in U for 25 cycles)<5% U(>95% dip in U for 5 s)

<5% U(>95% dip in U for 5 s)Note 2: Self recovers

30 A/m30 A/mЧестота на захранване
Магнитно поле (50/60 Hz)
IEC 61000-4-8Лампата за втвърдяване се захранва с батерии и неможе да се свърже към захранването с AC MAINS.
Тъй като светлината за втвърдяване се захранва сбатерии, тя не е обект на електрически преходни
процеси, пренапрежения, пропадания нанапрежение, къси паузи, прекъсвания или променив захранването с AC MAINS.
Допълнителното зарядно устройство за батерии
може да бъде предмет на горното, но е отделно ине е от решаващо значение за работата на VALO
Cordless curing light.
Ако напрежението на батерията на лампата законсервиране падне до 4VDC, устройството няма дапозволи работа. Светлината за втвърдяване VALOще се изключи. Когато се поставят нови батериии се възстановят правилните нива на мощност,светлината за втвърдяване ще се рестартира ище се върне в същото състояние преди загубатана мощност. Лампата за втвърдяване ще сесамовъзстанови в случай на загуба на мощност.
Магнитните полета с честотна мощност трябвада бъдат на нива, характерни за типично
местоположение в типична, жилищна, домашна
здравна, търговска, болнична или военна среда.

Бележка 1: Лампата за втвърдяване не е снабдена с никакви портове или достъпни I / O линии.
128

129

Ръководство и Декларация за производство на електромагнитни имунитети за системи за поддържане на живота
Светлината за втвърдяване е предназначена за използване в електромагнитната среда, посочена по-долу. Клиентът или потребителят трябва да се уверят, четой се използва в такава среда.
Тест IMMUNITY

IEC 60601 ниво наизпитване

Ниво на съответствие Ръководство за електромагнитна среда

Провеждане RF

3 Vrms3 VrmsIEC 61000-4-6

150 kHz до 80 MHz

150 kHz до 80 MHzПреносимо и мобилно радиочестотно комуникационно оборудване не трябва да сеизползва по-близо до която и да е част от светлината за втвърдяване, включително
кабелите, отколкото препоръчаното разстояние за разделяне, изчислено отуравнението, приложимо към честотата на предавателя.
Препоръчително разстояние за разделяне:80 MHz to 800 MHzИзлъчена RF

3V/m

3V/m

800 MHz до 2,5 GHz
P е максималната изходна мощност на предавателя във ватове (W) според
производителя на предавателя и d е препоръчителното разстояние в метри (m).
Силата на полето от фиксирани радиочестотни предаватели, определена отелектромагнитно изследване на мястото, трябва да бъде по-малка от нивото насъответствие във всеки честотен диапазонb.

IEC 61000-4-3

80 MHz до 2,5 GHz

80 MHz до 2,5 GHzМоже да възникнат смущения в близост до оборудване, маркирано със следния
символ:Декларация за ориентиране и производство за препоръчителни разстояния за разделяне между портативно и мобилно RF комуникационно оборудване и
VALO акумулаторна светлина за втвърдяване
Светлината за втвърдяване е предназначена за използване в електромагнитна среда, в която се контролират излъчвани RF смущения. Потребителят навтвърдяващата светлина може да помогне за предотвратяване на електромагнитни смущения чрез поддържане на минимално разстояние между преносимои мобилно RF комуникационно оборудване (предаватели) и светлината за втвърдяване, както се препоръчва по-долу, в съответствие с максималната изходна
мощност на комуникационното оборудване.

Номинална максимална изходна
мощностна предавателя
(P в ватове)Разстояние на разделяне според честотата на предавателя
(метра)150 kHz – 80 MHz

80 MHz – 800 MHz

800 MHz – 2,5 GHz

0,01

0,12 метра

0,035 метра

0,07 метра

0,10,37 метра

0,11 метра

0,22 метра

1

1,7 метра

0,35 метра

0,7 метра

10

3,7 метра

1,11 метра

2,22 метра

100

11,7 метра

3,5 метра

7,0 метра

Светлината за втвърдяване е тествана в съответствие с IEC 60601-1-2: 2014 и е преминала под силата на излъченото поле от 10 V / m между 80 MHz до 2,5
GHz. Стойността на 3Vrms съответства на V1 и стойността 10 V / m съответства на E1 във формулите по-горе.
За предаватели с максимална изходна мощност, които не са изброени по-горе, препоръчителното разстояние d в метри (m) може да бъде оценено, като сеизползва уравнението, приложимо за честотата на предавателя, където P е максималната мощност на изходната мощност на предавателя във ватове ( W)според производителя на предавателя.
ЗАБЕЛЕЖКА 1: При 80 MHz и 800 MHz се прилага разделителното разстояние за по-високия честотен диапазон.
ЗАБЕЛЕЖКА 2 Тези указания може да не се прилагат във всички ситуации. Електромагнитното разпространение се влияе от абсорбцията и отражението отструктури, обекти и хора.

ЗАБЕЛЕЖКА 1: При 80 MHz и 800 MHz се прилага по-високият честотен диапазон.
ЗАБЕЛЕЖКА 2 Тези указания може да не се прилагат във всички ситуации. Електромагнитното разпространение се влияе от абсорбцията и отражението отструктури, обекти и хора.
Сила на полето от фиксирани предаватели, като базови станции за радио (клетъчни / безжични) телефони и земни мобилни радиостанции, аматьорско
радио, AM и FM радиопредавания и телевизионни предавания, не могат да се предскажат теоретично с точност. За да се оцени електромагнитната среда,дължаща се на фиксирани радиочестотни предаватели, трябва да се обмисли електромагнитно изследване на мястото. Ако измерената напрегнатост наполето в мястото, където се използва светлината за втвърдяване, надвишава приложимото ниво на радиочестотно съответствие по-горе, акумулаторната
светлина VALO трябва да се наблюдава, за да се провери нормалната работа. Ако се наблюдават ненормални характеристики, може да са необходими
допълнителни мерки, като преориентиране или преместване на светлината за втвърдяване.
В честотния диапазон от 150 kHz до 80 MHz силата на полето трябва да бъде по-малка от 3 V / m.

130

131

VALO™ GRANDHRSVJETLO ZA STVRDNJAVANJE
Opis proizvoda

Sa svojim širokopojasnim spektrom, VALO Grand svjetiljka za polimeriziranje dizajnirana je za polimeriziranje svih svjetlosno polimerizirajućih proizvoda u rasponu valnih duljina od
385-515 nm prema ISO 10650. VALO Grand svjetiljka za polimeriziranje koristi Ultradent VALO punjive baterije i punjač baterija. Svjetiljka za polimeriziranje dizajnirana je za postavljanje ustandardni nosač stomatološke jedinice ili se može montirati po narudžbi pomoću nosača koji je uključen u komplet.

Komponente VALO Grand proizvoda:
•
•
•
•
•
•

Fotografija

1) Napunite baterije prije korištenja svjetiljke za polimeriziranje (pogledajte odjeljak Održavanje baterija).
2) Prije svake uporabe, stavite novu zaštitnu navlaku preko svjetiljke za polimeriziranje, smanjujući nabore na leći za najbolje rezultate.

VALO Bežični štitnik svjetla:

VALO Bežični štitnik svjetla je ovalnog oblika, može se okretati radi maksimalnog korištenja i koristiti s prozirnim zaštitnim rukavcem.

Uporaba
Gumbi i svjetla

LegendaSvjetla mjerača vremena
Mala baterija - Pokazatelj
Svjetlosni signal / način rada
Gumb za mjerenje vremena/načina rada(Vrh)

Za sve opisane proizvode pažljivo pročitajte i razumijte sve upute i podatke o sigurnosti prije korištenja.

Indikacije za uporabu / namjena

Izvor osvjetljenja za stvrdnjavanje foto-aktiviranih dentalnih restorativnih materijala i ljepila.

Upozorenja i mjere opreza
Grupa rizika 2
OPREZ, ovaj proizvod emitira UV zrake. Izlaganje može uzrokovati nadraživanje očiju ili kože. Koristite odgovarajuću zaštitu.
OPREZ Može doći do opasnog optičkog zračenja ovog proizvoda. Ne gledajte u radnu lampu. Može biti štetno za oči.
• NEMOJTE gledati izravno u rezultirajuće svjetlo. Pacijent, kliničar i asistenti uvijek trebaju nositi UV zaštitu za oči boje jantara kada se koristi svjetiljka za polimeriziranje.
• Kako bi se spriječio rizik od električnog udara, nisu dopuštene nikakve izmjene ove opreme. Koristite samo priložene Ultradent VALO napajanje i adaptere utikača. Ako su tekomponente oštećene, nemojte ih koristiti i nazovite Ultradent službu za korisnike kako biste naručili zamjenu.
• Prijenosna RF komunikacijska oprema može smanjiti performanse ako se koristi bliže od 30 cm (12 in.).
• Koristite samo odobreni pribor, kabele i izvore napajanja kako biste spriječili neispravan rad, povećane elektromagnetske emisije ili smanjenu elektromagnetsku otpornost(pogledajte odjeljak Elektromagnetske emisije).
• Kako biste izbjegli rizik od električnog požara povezanog s rukovanjem baterijama:⸰ NEMOJTE miješati punjive baterije s nepunjivim baterijama ili drugim vrstama baterija.
⸰ NEMOJTE pokušavati puniti baterije koje se ne mogu puniti. Punjenje nepunjivih baterija može rezultirati ozljedama ili oštećenjem imovine.
⸰ NEMOJTE autoklavirati niti prskati baterije, kontakte na baterijama, punjač ili AC adapter bilo kojom vrstom tekućine. Ako se na kontaktima punjača baterija pojavi korozija,nazovite Ultradent službu za korisnike kako biste naručili zamjenu.
⸰ NEMOJTE puniti baterije u blizini zapaljivih materijala.
⸰ NEMOJTE držati punjač u kliničkoj ordinaciji.
• Kako biste izbjegli rizik od ozljeda, NEMOJTE koristiti baterije koje su korodirale (zahrđale), ulubljene, ispuštaju miris ili tekućinu, imaju poderan omot ili im on nedostaje, ili su na neki
drugi način oštećene. Nazovite Ultradent službu za korisnike kako biste naručili zamjenske baterije.
• Koristite samo preporučene baterije. Alternativne baterije mogu uzrokovati kvar.
• Kako biste spriječili rizik od toplinske iritacije ili ozljede, izbjegavajte uzastopne cikluse polimeriziranja i ne izlažite oralna meka tkiva u neposrednoj blizini dulje od 10 sekundi u bilo
kojem načinu rada. Ako su potrebna dulja vremena polimeriziranja, koristite višestruke cikluse polimeriziranja s razdobljima odmora između ciklusa ili koristite proizvod za dvostruko
polimeriziranje kako biste izbjegli zagrijavanje mekog tkiva.
• Budite oprezni pri liječenju pacijenata koji pate od neželjenih fotobioloških reakcija ili osjetljivosti, pacijenata koji su podvrgnuti kemoterapiji ili pacijenata koji se liječe
fotosenzibilizirajućim lijekovima.
• Ova jedinica može biti osjetljiva na jaka magnetska ili statička električna polja, koja mogu poremetiti programiranje. Ako sumnjate da se to dogodilo, isključite uređaj na trenutak, azatim ga ponovno uključite u utičnicu.
• NEMOJTE brisati svjetiljku za polimeriziranje kaustičnim ili abrazivnim sredstvima za čišćenje, autoklavirati niti uranjati u bilo koju vrstu ultrazvučne kupke, dezinficijensa, otopine začišćenje ili tekućine. Nepoštivanje priloženih uputa za obradu može učiniti svjetiljku za polimeriziranje nefunkcionalnom.
132

Postupne upute
Priprema• Kako bi se spriječila unakrsna kontaminacija i spriječilo prijanjanje dentalnog kompozitnog materijala na površinu leće i tijela štapića, pri svakoj uporabi preko VALO svjetiljke za
polimeriziranje mora se koristiti zaštitna navlaka koju je odobrio Ultradent. Zaštitne navlake namijenjene su za uporabu kod jednog pacijenta.

Gumb(-) za napajanje(Gore ili dolje)(Dno)

Kako biste izbjegli oštećenje opreme, NEMOJTE umetati prste, instrumente ili druge predmete u odjeljak za baterije svjetiljke za polimeriziranje.
Kako biste izbjegli oštećenje opreme, NEMOJTE pokušavati očistiti zlatne kontakte ili bilo koji dio odjeljka za baterije. Nazovite Ultradent službu za korisnike ako postoji nedoumica.
Kako bi se spriječio rizik od unakrsne kontaminacije, zaštitne navlake namijenjene su samo jednom pacijentu.
Kako biste smanjili rizik od korozije, uklonite zaštitni omotač nakon uporabe.
Kako biste smanjili rizik od nedovoljno polimeriziranih smola, nemojte koristiti svjetiljku za polimeriziranje ako je leća oštećena.

Napomena:

1 – VALO Grand svjetiljka za polimeriziranje
4 – Ultradent VALO punjive baterije
1 – Ultradent VALO punjač baterija s medicinskim strujnim adapterom od 12 VDC AC
1 – VALO Zaštitna navlaka paket uzoraka
1 – VALO Grand svjetlosni štit1 – Nosač za površinsku montažu svjetiljke za polimeriziranje s dvostrukom ljepljivom trakom

Pregled kontrola:

•
•
•
•
•

1) Svaki način napajanja koristi se za stvrdnjavanje dentalnih materijala s foto inicijatorima. Za preporučena vremena stvrdnjavanja pogledajte Vodič za brzi način rada.
NAPOMENA: Svjetlo za stvrdnjavanje programirano je da se prebacuje sa standardnog napajanja do visoke snage u način rada ekstra napajanja. Primjerice, za promjenu iz standardnog
načina rada u ekstra način, potrebno je najprije pokrenuti način rada visoke snage, a zatim u način rada ekstra napajanja.
2) Svjetlo za stvrdnjavanje pohranjuje posljednji korišteni vremenski interval i način rada i vratit će se na ovaj način rada kad se promijeni način rada ili ako se baterije uklone.

RadNAČIN RADA: Standardni način napajanja

Vremenski intervali: 5, 10, 15, i 20 sekundi.
• Ovo je zadani način rada svjetla za stvrdnjavanje kad ga INCIJALNO uključite. Svjetlo načina rada/statusa će biti zeleno, a četiri zelena svjetla će svijetliti, što označava standardni način
rada.
• Za promjenu vremenskih intervala, brzo pritisnite gumb za promjenu vremena/načina rada.
• Pritisnite gumb za uključivanje i isključivanje. Za zaustavljanje sušenja prije završetka vremenskog intervala ponovno pritisnite gumb za uključivanje.

NAČIN RADA: Način rada visoke snage

Vremenski intervali: 1, 2, 3, i 4 sekundi.
• U načinu rada za standardno napajanje pritisnite i držite gumb za promjenu vremena/načina rada 2 sekunde i otpustite. Svjetlo načina rada/statusa bit će narančasto, a četiri zelena
svjetla vremena će zasvijetliti i treptati, što označava način rada visoke snage.
• Za promjenu vremenskih intervala, brzo pritisnite gumb za promjenu vremena/načina rada.
• Pritisnite gumb za uključivanje za početak stvrdnjavanja. Za zaustavljanje stvrdnjavanja prije završetka vremenskog intervala ponovno pritisnite gumb za uključivanje/isključivanje.
• Za povratak na standardni način rada, pritisnite i držite gumb za promjenu vremena/načina rada 2 sekunde i pustite, to će se prebaciti na način rada ekstra napajanja. Ponovno
pritisnite i držite još 2 sekunde i otpustite. Svjetlo načina rada/statusa bit će zeleno, a četiri zelena svjetla vremena će svijetliti, što označava standardni način rada.

NAČIN RADA: Način rada ekstra napajanja

VREMENSKI INTERVAL: Samo 3 sekunde (Napomena: Način rada ekstra napajanja ima 2 sekunde sigurnosne odgode na kraju svakog ciklusa stvrdnjavanja da ograniči zagrijanje tijekom
uzastopnog sušenja. Na kraju odgode, zvučni signal označava da je jedinica spremna za nastavak korištenja).
• U standardnom načinu rada pritisnite gumb za promjenu vremena/načina rada na 2 sekunde, otpustite, ponovno pritisnite i držite 2 sekunde i otpustite. Svjetlo načina rada/statusa
bit će narančasto i zatreperiti, a tri zelena svjetla vremena će zasvijetliti i zatreperiti, što označava način rada ekstra napajanja.
• Pritisnite gumb za uključivanje i isključivanje. Za zaustavljanje sušenja prije završetka vremenskog intervala ponovno pritisnite gumb za uključivanje.
• Za povratak na standardni način rada, pritisnite i držite gumb za promjenu vremena/način rada 2 sekunde i otpustite. Svjetlo načina rada/statusa bit će zeleno, a zelena svjetla
vremena će se osvijetliti, što označava standardni način rada.

Stanje mirovanja: Svjetlo za stvrdnjavanje prelazi u stanje mirovanja nakon 60 sekundi neaktivnosti, što je označeno laganim treptanjem svjetla načina rada/statusa. Podizanje ili
dodirivanje uređaja će probuditi svjetlo za stvrdnjavanje i automatski ga vratiti na zadnju postavku. Kako biste maksimalno produžili vijek trajanja baterije, ostavite svjetlo za stvrdnjavanje
neometano kada se ne koristi.

Čišćenje

1) Iskorištene zaštitne rukavce odbacite u standardni otpad nakon svakog pacijenta.
2) Pogledajte odjeljak Obrada.

Upute za montažu nosača

1) Nosač treba montirati na ravnu, bezuljnu površinu.
2) Očistite površinu alkoholom.
3) Skinite poleđinu s ljepljive trake nosača.
4) Postavite nosač tako da se svjetlo za stvrdnjavanje podigne prema gore kada ga uklonite. Čvrsto pritisnite na mjesto.

133

Vodič za brzi način rada:
Standardna snaga
1000 mW/cm2

Način

Visoka snaga napajanja
1600 mW/cm2

Ekstra Power
3200 mW/cm2

Gumb za uključivanje
LED diode za način rada/
vremena
Gumbi vremena
Opcije vremena

5s

10s

15s

20s

1s

2s

3s

4s

Samo 3s

Promjena vremena

Brzo pritisnite i otpustite gumb za promjenu vremena za kretanje kroz vremenske opcije.

Promjena načina rada

Pritisnite i držite tipku za vremenske postavke u trajanju od 2 sekunde i otpustite. Svjetlo za stvrdnjavanje prelazi u sljedeći
način rada.

LegendaPuna LED svjetla

Treptajuća LED svjetla

Vodič za brzo sušenje:
Način rada za
polimeriziranje

Snaga* (mW)Ozračenje* (mW/cm2)

Ukupno vrijeme izloženosti (sekunde)

Energija** (džuli)Standard

970

900

5/10/15/20

4,8 J/9,7 J/14,5 J/19,4 J

Snaga High Power Plus

1.620

1.500

1/2/3/4

1,6 J/3,2 J/4,8 J/6,5 J

Snaga Xtra

2.260

2.100

3

6,8J

*Nazivni izlaz na udaljenosti od 2 mm od vrha leće do gornje površine kompozita.
**Vrijednosti u odjeljku ukupne energije po ciklusu (džuli) zaokružene su na najbližu deseticu.

Zamijenite baterije• Niska razina baterije: sporo treperi• Isključivanje: 3 zvučna signala,treperenje• Zabranjuje rad

Upozorenje o temperaturi

Upozorenje o usluzi

LED upozorenje

Pustite da se ohladi

Nazovite službu za korisnike za
popravakNazovite službu za korisnike za
popravak• 3 zvučna signala• Polagano treperenje• Zabranjuje rad• Bez zvuka• Treperi svake 2 sekunde• Omogućuje rad

Kako puniti baterije:
1) Punjač i baterije stavite na mjesto gdje neće doći u kontakt s tekućinama ili sprejom za dezinfekciju.
2) Umetnite baterije u punjač tako da pozitivni (+) kraj bude usmjeren prema indikatorskim lampicama na punjaču.
3) Zelene lampice pokazuju da su baterije spremne za upotrebu.
4) Za punjenje baterija potrebno je 1-3 sata. Ostavite baterije u punjaču dok ne budu spremne za upotrebu.
NAPOMENA: Ako crvena lampica na punjaču ne svijetli zeleno kada su se baterije punile dulje od tri sata, baterija je možda degradirana (oštećena) i ne može se puniti. Isprobajte novu
bateriju ili nazovite Ultradent službu za korisnike kako biste naručili novi komplet punjivih baterija.
Kako promijeniti/umetnuti baterije:
1) Uklonite stražnji poklopac okretanjem u smjeru suprotnom od kazaljke na satu za četvrtinu kruga.
2) Izvadite baterije.
3) Umetnite nove baterije prvo s pozitivnom (+) stranom.
4) Ponovno pričvrstite stražnji čep poravnavanjem i nježnim guranjem okrećući u smjeru kazaljke na satu. Čep će kliknuti kada je potpuno pričvršćen.
5) Jedinica je spremna za upotrebu.
Slabe baterije: Svjetiljka za polimeriziranje signalizira korisniku da je vrijeme za promjenu baterija kada lampica indikatora za slabu bateriju trepti crveno. Ako napunjenost baterije
postane preniska, oglasit će se zvučna upozorenja od 3 zvučna signala i svjetiljka za polimeriziranje neće dopustiti daljnji rad dok se baterije ne napune ili umetnu nove baterije.
(Pogledajte Kratki vodič za upozorenje)
Popravak od strane proizvođača
1) Popravke smije obavljati samo ovlašteno servisno osoblje. Ultradent treba servisnom osoblju osigurati dokumentaciju za obavljanje popravaka.
2) Kada šaljete jedinice na popravak, servis ili kalibraciju, uvijek uklonite baterije iz svjetiljke za polimeriziranje i punjača. Zamotajte baterije, punjač, adapter i svjetiljku za polimeriziranje
zasebno u povratnu kutiju.
3) Pošaljite baterije u skladu s lokalnim propisima.
Jamstvo
Ultradent Products, Inc. ("Ultradent") jamči da će ovaj proizvod, tijekom razdoblja od 5 godina od datuma kupnje, kada se njime rukuje u skladu s uputama za uporabu priloženim uz
proizvod, (i) u svim materijalnim aspektima biti usklađen s specifikacije navedene u Ultradent dokumentaciji koja prati proizvod; i (ii) biti bez nedostataka u materijalu i izradi.
Ovo ograničeno jamstvo nije prenosivo i odnosi se isključivo na prvobitnog kupca i ne odnosi se na buduće vlasnike proizvoda. Ovo ograničeno jamstvo ne pokriva druge komponentekao što su, ali ne ograničavajući se na, baterije, punjače, adaptere ili adaptivne leće. Ovo ograničeno jamstvo je nevažeće ako proizvod pokvari ili je oštećen zbog nemara, zlouporabe,
pogrešne uporabe, nesreće, modifikacije, petljanja, izmjene ili nepoštivanja primjenjivih uputa za uporabu. Samo u svrhu primjera, proizvod koji je ispušten i oštećen nije pokriven ovim
jamstvom. Za ispunjavanje uvjeta za ovo ograničeno jamstvo, dokaz o kupnji (npr. račun ili slična dokumentacija) mora se dostaviti Ultradentu zajedno s neispravnim proizvodom.

Vodič za brzo upozorenje:
Upozorenje o razini napajanja

Očistite VALO Cordless svjetlosni štit bilo kojim sredstvom za dezinfekciju površina. NEMOJTE autoklavirati.
PopravakOdržavanje koje provodi korisnik
1) Koristite zaštitnu navlaku kako biste spriječili lijepljenje zubnog kompozita na površinu leće. Ako je potrebno, koristite zubarski instrument od plastike ili nehrđajućeg čelika kako bistepažljivo uklonili sav zalijepljeni kompozit. Nemojte koristiti alate koji će oštetiti leću.
2) Svjetlomjeri se uvelike razlikuju i dizajnirani su za specifične vrhove i leće svjetlovoda. Ultradent preporučuje rutinsku provjeru izlaza u načinu rada Standard Power. NAPOMENA: pravi
numerički izlaz bit će iskrivljen zbog nepreciznosti uobičajenih svjetlomjera i prilagođenog LED paketa u svjetiljki za polimeriziranjeOdržavanje baterije
Punjenje i mijenjanje baterija
Svjetiljka za polimeriziranje dolazi s 4 punjive litij željezo fosfatne baterije. Ultradent preporuča korištenje samo VALO punjivih baterija.

• Neprekidna 3 zvučna signala• Jedna brza bljeskalica• Zabranjuje rad

Neispravan proizvod koji ispunjava ovdje navedene uvjete jamstva će, prema isključivom nahođenju Ultradenta, biti popravljen ili zamijenjen. Ni u kojem slučaju Ultradentova
odgovornost za proizvod neće premašiti kupovnu cijenu koju je platio kupac. Ultradent ni pod kojim okolnostima neće biti odgovoran za bilo kakvu neizravnu, slučajnu, predviđenu,nepredviđenu, posebnu ili posljedičnu štetu koja proizlazi iz ili u vezi s uporabom ovog proizvoda.

ObradaOvaj proizvod je namijenjen za korištenje sa zaštitnom navlakom. Vidi gore za održavanje i opće čišćenje.

Skladištenje i zbrinjavanje

Ako spremite svjetlo za stvrdnjavanje dulje od 2 tjedna ili pakirate za putovanje, uvijek izvadite baterije. Ako se baterije u uređaju ostavljaju dulje vrijeme bez punjenja, one mogu postati
nefunkcionalne ili se ne mogu više napuniti. Ne ostavljajte baterije na temperaturama iznad 40 °C (14°F) ili na izravnom sunčevom svjetlu.
Stvrdnjavanje svjetla Skladištenje i transport:• Temperatura: + 10 ° C do + 40 ° C (+ 50 ° F do + 104 ° F)
• Relativna vlažnost: 10% do 95%• Ambijentalni tlak: 500 hPa do 1060 hPa
Prilikom zbrinjavanja elektroničkog otpada (tj. Svjetla za stvrdnjavanje, punjača, baterija i izvora napajanja), slijedite lokalne smjernice za otpad i recikliranje.

Održavanje

Općenito čišćenje svjetiljke za polimeriziranje
Nakon svake uporabe navlažite gazu ili meku krpu odobrenim sredstvom za dezinfekciju površina i obrišite površinu i leću. Neovlaštena sredstva za čišćenje mogu oštetiti svjetiljku za
polimeriziranje.

Tehnička razmatranja
Pribor

PRIHVATLJIVA SREDSTVA ZA ČIŠĆENJE:• 70% izopropilni alkohol• 70% etanol
Općenito čišćenje svjetlosnog štita:
134

135

VALO zaštitni rukavci

Tehnički podaci/podaci

CE informacije

MDSS GmbH
Schiffgraben 4130175 Hanover
Njemačka

Proizveo:
TIDI Products, LLC.
570 Enterprise Drive
Neenah, WI 54956
Napravljeno u americi

Učinkovite valne duljine za stvrdnjavanje kompozita:
Distribuira:Ultradent Products Inc
505 West Ultradent Drive (10200 Jug)Južni Jordan, UT 84095
SADUČINKOVITO POJASANJE VALNIH DUŽINA KOMPOZITA(Koristivi raspon valnih duljina: 385-515 nm)
SPEKTRALNA SNAGA ZRAČENJA (mW/nm)

ArtiklVALO Bežični štitnik svjetla

VALO punjač

VALO baterije

Učinkovito nižepojas valnih duljina
395-415 nm

Efektivni pojas gornje
valne duljine
440-480 nm

Snaga Xtra
Snaga High PowerStandard

VALITI ENGTH (nm)Atribut

Informacije/Specifikacija

Leće

Promjer 11,7 mm

Raspon valnih
duljina• Raspon valnih duljina: 385 - 515 nm• Vršne valne duljine: 395 - 415 nm i 440 - 480 nm
Nazivna usporedna tablica ozračenja
Mjerni instrument

Tablica intenziteta
svjetla136‡ Gigahertz analizator spektra
Ozračenje

Ukupna snaga

Otvor mjerača

15 mm

15 mm

Standardna snaga(±10%)

900 mW/cm²

970 mW

Visoka snaga
napajanja (±10%)1500 mW/cm²

1615 mW

Xtra snaga (±10%)2100 mW/cm²

Radiant Exitance će varirati ovisno omogućnostima instrumenta, metodi mjerenja ipoložaju svjetilljke.
‡ Radiant Exitance je u skladu s ISO
10650 kada se mjeri analizatorom spektra
Gigahertz-Optik.

2260 mW

VALO Grand svjetlo
za stvrdnjavanje

Ocjene: IEC 60601-1 (Sigurnost), IEC 60601-1-2(EMC)

Visina:• S baterijama: 6 oz. (190 grama)• Bez baterija: 5 oz. (150 grama)
Dimenzija: (8 x 1,28 x 1,06) inča, (203 x 32,5 x 27) mm

Napajanje punjača

Izlaz - 12VDC na 500mA
Ulaz - 100VAC do 240VAC
Ultradent P/N 5930 Napajanje punjača smeđunarodnim utičnicama

Ocjene: IEC 60601-1 (Sigurnost)
Duljina kabela - 6 stopa (1,8 metara)
VALO punjač ima napajanje medicinskog razreda klase II i osigurava
izolaciju od GLAVNOG IZVORA napajanja

VALO punjač

VALO 3.6VDC Pametni punjač za baterije od litij-željezo-fosfata:• Automatsko isključivanje kada je potpuno napunjeno• Automatsko otkrivanje neispravnih baterija• Zaštite: Termička, preopterećenje, kratki spoj, obrnuti polaritet
⸰ Crveni LED – Punjenje
⸰ Zeleni LED – Prazno ili u potpunosti napunjeno
⸰ LED isključen - kratki spoj
• Vrijeme punjenja: 1 - 3 sata
Ocjena: CE, WEEE
137

Atribut

Informacije/Specifikacija

VALO baterije

Punjive: Siguran litij-željezo-fosfat (LiFePO4) RCR123A• Radni napon: 3.2VDC
Ocjene: UL, CE, RoHS, WEEE

Radni uvjeti

Temperatura: +10°C to +32°C (+50°F to +90°F)
Relativna vlažnost: 10 % do 95 %Ambijentalni tlak: 700 hPa do 1060 hPa

Radni ciklus

Ostale informacije
Smjernice i izjava proizvođača o elektromagnetskim emisijama
Svjetlo za stvrdnjavanje namijenjeno je za uporabu u elektromagnetskom okruženju navedenom u nastavku. Kupac ili korisnik treba osigurati da se koristi u takvom
okruženju.

Svjetlo za stvrdnjavanje dizajnirano je za kratkotrajni rad. Pri maksimalnoj temperaturi okoline (32 °C) 1 minuta UKLJUČUJUĆI
povratni ciklus, 30 minuta ISKLJUČENO (razdoblje hlađenja).

Uklanjanje problema
Ako rješenja navedena u nastavku ne otklone problem, nazovite Ultradent na 800.552.5512. Izvan Sjedinjenih Američkih Država, nazovite svog distributera ili trgovca
dentalnom opremom.
ProblemMoguća rješenja

Ispitivanje emisija

Usklađenost

Elektromagnetsko okruženje - smjernice

RF emisije
CISPR 11

Skupina 1

Svjetlo za stvrdnjavanje koristi baterije i ne utječe na EMI, RF ili prenaponsku zaštitu.

RF emisija
CISPR 11

Klasa B

Svjetlo za stvrdnjavanje koristi električnu i elektromagnetsku energiju samo za svoje unutarnje
funkcije. Stoga, sve su RF emisije vrlo niske i vjerojatno neće uzrokovati smetnje obližnjoj
elektroničkoj opremi.

Harmonijske emisije
IEC 61000-3-2

N/A

Ispitivanje harmonijskih emisija i fluktuacija napona nije primjenjivo na svjetlo za stvrdnjavanje jer
se napaja baterijom.

Fluktuacije napona/treperenje
IEC 61000-3-3

N/A

Svjetlo za stvrdnjavanje pogodno je za upotrebu u svim objektima, uključujući domaće objekte i
one koji su izravno povezani s javnom niskonaponskom mrežom za napajanje, koja opskrbljuje
zgrade za kućnu uporabu.

Svjetlo se neće upaliti

1) Pritisnite gumb za promjenu vremena/načina rada ili gumb za uključivanje da biste pokrenuli uređaj iz načina uštede
energije.
2) Provjerite crveni indikator niskog stanja baterije za stanje napunjenosti baterije.
3) Provjerite da li su nove baterije ispravno umetnute u uređaj.
4) Ako crvene i žute LED svjetla upozorenja trepere, to znači da je svjetlo za stvrdnjavanje dostiglo sigurnosnu granicu
unutarnje temperature. Pustite da se svjetlo za stvrdnjavanje ohladi u trajanju od 10 minuta ili upotrijebite hladan ručnik
kako biste uređaj brzo ohladili.
5) Ako crveno LED upozorenje treperi i stalno se oglašava, nazovite Ultradent službu za korisnike zbog popravka.

Svjetlo se ne ostaje upaljeno u željenom
vremenu.

1) Provjerite način rada i vremenska svjetla za ispravan unos vremena.
2) Provjerite indikator niskog stanja napunjenosti baterije.
3) Provjerite da li su nove baterije ispravno umetnute u uređaj.

Ispitivanje IMUNITETA

IEC 60601 ispitna razina

Razina usklađenosti

Vodič za elektromagnetsko okruženje

Elektrostatičko pražnjenje (ESD)
IEC 61000-4-2± 8 kV kontakt± 15 kV zraka

Svjetlo ne stvrdnjava pravilno smolu

1) Provjerite leću za naslage polimerizirane smole/kompozita.
2) Koristeći odgovarajuću zaštitu za oči od UV zračenja u boji jantara, provjerite pravilan rad LED svjetla.
3) Provjerite razinu snage pomoću mjerača svjetla. Ako koristite mjerač svjetlosti, Ultradent preporučuje provjeravanje
svjetla za stvrdnjavanje standardnom načinu rada.
NAPOMENA: Pravi brojčani izlaz će biti iskrivljen zbog netočnosti uobičajenih mjerača svjetlosti i prilagođenog LED paketa koji
koristi svjetlo za stvrdnjavanje. Svjetlomjeri se uvelike razlikuju i namijenjeni su za određene svjetlosne vodilice i leće.
4) Provjerite datum isteka na polimerizirajućoj smoli.
5) Pridržavajte se pravilne tehnike prema preporukama proizvođača.

± 8 kV kontakt± 15 kV zraka

Fizičko okruženje treba ograničiti na sljedeće:
1) IP kod: IP20
2) Ne uranjajte u tekućinu.
3) Ne koristite u blizini zapaljivog plina. Jedinica je
Non-APG i Non-AP.
4) Raspon vlažnosti spremnika: 10 % - 95 %5) Raspon temperature spremnika: 10 °C - 40 °C

Brzi električni tranzijenti/erupcija
IEC 61000-4-4± 2 kV za vodove napajanja± 1 kV za ulazne / izlazne linije± 2 kV za vodove napajanja
Napomena 1: Svjetlo za
stvrdnjavanje nema ulaza I / 0

Svjetlo za stvrdnjavanje pokreće se na baterije i nemože se povezati s GLAVNIM izvorom AC napajanja.

Udar
IEC 61000-4-5± 1 kV linija do linije± 2 kV linija do uzemljenja± 1 kV linija do linije± 2 kV linija do uzemljenja

Napon, padovi, kratki spojevi,prekidi i varijacije na ulaznim
linijama napajanja<5% U(> 95 % pad u ciklusu U za 0,5)

<5 % U(> 95 % pad u ciklusu U za 0,5)

40 % U(60 % umočiti u U za 5 ciklusa)

40 % U(60 % umočiti u U za 5 ciklusa)

70 % U(30 % umočiti u U za 25 ciklusa)

70 % U(30% umočiti u U za 25 ciklusa)<5 % U(> 95 % uronite u U za 5 s)

<5 % U(> 95 % uronite u U za 5 s)
Napomena 2: Sam se oporavlja

30 A/m30 A/mBaterije se neće puniti

1) Provjerite da li su baterije umetnute u punjač u ispravnom smjeru i pustite da se pune 1-3 sata.
2) Ako se crvena svjetla na punjaču ne promijene u zeleno, nazovite službu za korisnike Ultradent i naručite zamjenskebaterije i/ili punjač.
3) Ako se ne vide zelena ni crvena svjetla na punjaču, nazovite Ultradent službu za korisnike kako biste naručili ili zamijenili
punjač i/ili AC adapter.

Punjač ne puni baterije

1) Provjerite da li je punjač priključen, a AC adapter priključen u radnu utičnicu.
2) Ako zelena ili crvena svjetla na punjaču nisu vidljiva, nazovite Ultradent službu za korisnike za novi punjač i/ili AC adapter.

Nije moguće promijeniti način ili
vremenske intervale1) Držite pritisnute gumbe za promjenu vremena/načina rada i gumb za uključivanje uređaja sve dok niz zvučnih signala neoznači otključavanje svjetla za stvrdnjavanje.

Smjernice i izjava proizvođača o elektromagnetskom imunitetu
Svjetlo za stvrdnjavanje namijenjeno je za uporabu u elektromagnetskom okruženju navedenom u nastavku. Kupac ili korisnik treba osigurati da se koristi u takvom
okruženju.

IEC 61000-4-11

Frekvencija napajanja(50/60 Hz) magnetsko polje
IEC 61000-4-8

Budući da se svjetlo za stvrdnjavanje napaja pomoću
baterija, ono ne podliježe električnim tranzijentima,
prenaponima, padovima napona, kratkim spojevima,
prekidima ili varijacijama na GLAVNOM izvoru AC
napajanja.
Dodatni punjač može podlijegati gore navedenom, ali
je odvojen od i nije kritičan za rad VALO Bežični svjetla
za stvrdnjavanje.
Ako napon baterije svjetla za stvrdnjavanje padne
na 4VDC, uređaj neće dopustiti rad. Svjetlo za
Stvrdnjavanje VALO Bežični će se isključiti- Kada seumetnu nove baterije i vrate odgovarajuće razine
napajanja, svjetlo će se ponovno pokrenuti i vratiti
u isto stanje kao i prije gubitka napajanja. Svjetlo za
stvrdnjavanje će se samostalno oporaviti u slučaju
gubitka napajanja.
Power frequency magnetic fields should be atlevels characteristic of a typical location in a typical,
residential, home health care, commercial, hospital, or
military environment.

Napomena 1: Svjetlo za stvrdnjavanje nije opremljeno nikakvim ulazima ili bilo kojim dostupnim I / O linijama.

138

139

Smjernice i izjava proizvođača o elektromagnetskom imunitetu za sustave koji ne služe održavanju života

Smjernice i izjava proizvođača o preporučenoj udaljenosti razdvajanja između prijenosne i mobilne RF komunikacijske opreme i VALO Bežični svjetla za stvrdnjavanje

Svjetlo za stvrdnjavanje namijenjeno je za uporabu u elektromagnetskom okruženju navedenom u nastavku. Kupac ili korisnik treba osigurati da se koristi u takvom
okruženju.

Svjetlo za stvrdnjavanje namijenjeno je za uporabu u elektromagnetskom okruženju u kojem se kontroliraju smetnje RF zračenja. Korisnik svjetla za stvrdnjavanjemože spriječiti elektromagnetske smetnje održavanjem minimalne udaljenosti između prijenosne i mobilne RF komunikacijske opreme (odašiljači) i svjetla za
stvrdnjavanje u skladu s niže navedenom preporučenom maksimalnom izlaznom snagom komunikacijske opreme.

Ispitivanje IMUNITETA

IEC 60601 ispitna
razinaRazina usklađenosti

Provođenje RF

3 Vrms3 VrmsIEC 61000-4-6

150 kHz do 80 MHz

150 kHz do 80 MHz

Ozračeni RF

3 V/m

3 V/m

Vodič za elektromagnetsko okruženje
Prijenosna i mobilna RF komunikacijska oprema ne bi se trebala koristiti bliže bilo kojem
dijelu svjetla za stvrdnjavanje, uključujući kabele, od preporučene udaljenosti razdvajanja
izračunate pomoću jednadžbe koja se primjenjuje na frekvenciju odašiljača.
Preporučena udaljenost razdvajanja::

80 MHz do 800 MHz

800 MHz do 2,5 GHz
P je najveća izlazna snaga odašiljača u vatima (W) prema proizvođaču odašiljača, a d je
preporučena udaljenost razdvajanja u metrima (m).
Jačina polja fiksnih EF odašiljača, kako je utvrđeno elektromagnetskim ispitivanjem na
lokacijia, treba biti manja od razine usklađenosti u svakom frekvencijskom rasponub.

IEC 61000-4-3

80 MHz do 2,5 GHz

80 MHz do 2,5 GHz

Mogu se pojaviti smetnje u blizini opreme označene sljedećm simbolom:

Udaljenost razdvajanja prema frekvenciji odašiljača(metara)Nazivna maksimalna izlazna snaga
odašiljača(P u vatima)

150 kHz – 80 MHz

80 MHz – 800 MHz

800 MHz – 2,5 GHz

0,01

0,12 metara

0,035 metara

0,07 metara

0,10,37 metara

0,11 metara

0,22 metara

1

1,7 metara

0,35 metara

0,7 metara

10

3,7 metara

1,11 metara

2,22 metara

100

11,7 metara

3,5 metara

7,0 metara

Svjetlo za stvrdnjavanje testirano je u skladu s IEC 60601-1-2: 2014 i prošlo testiranje s jačinom polja zračenja od 10 V/m između 80 MHz i 2,5 GHz. Vrijednost 3Vrms
odgovara V1, a vrijednost 10 V/m odgovara E1 u gornjim formulama.
Za odašiljače s maksimalnom izlaznom snagom koja nije navedena gore, preporučena udaljenost razdvajanja d u metrima (m) može se procijeniti pomoću jednadžbe
koja se primjenjuje na frekvenciju odašiljača, gdje je P najveća snaga izlazne snage odašiljača u vatima ( W) prema proizvođaču predajnika.
NAPOMENA 1: Na 80 MHz i 800 MHz primjenjuje se udaljenost razdvajanja za viši frekvencijski raspon.
NAPOMENA 2: ove smjernice možda se neće primjenjivati u svim situacijama. Na elektromagnetsko širenje utječe apsorpcija i refleksija od struktura, objekata i ljudi.

NAPOMENA 1: pri 80 MHz i 800 MHz primjenjuje se viši frekvencijski raspon.
NAPOMENA 2: ove smjernice možda se neće primjenjivati u svim situacijama. Na elektromagnetsko širenje utječe apsorpcija i refleksija od struktura, objekata i ljudi.
Polja jakosti fiksnih odašiljača, kao što su bazne stanice za radio, (mobilne/bežične) telefone i kopnene mobilne radio uređaje, amaterski radio, AM i FM radioi TV emitiranje ne mogu se teoretski predvidjeti s točnošću. Da bi se procijenila elektromagnetska okolina zbog fiksnih RF odašiljača, treba izvesti ispitivanjeelektromagnetskog mjesta. Ako izmjerena jakost polja, na mjestu na kojem se koristi svjetlo za stvrdnjavanje, premašuje gore navedenu primjenjivu razinu
usklađenosti s RF, VALO Bežični svjetlo za stvrdnjavanje potrebno je promatrati kako bi provjerili pravilan rad. Ako primijetite neobične performanse, možda će biti
potrebne dodatne mjere, kao što je preusmjeravanje ili premještanje svjetla za stvrdnjavanje.
b
Iznad frekvencijskog raspona od 150 kHz do 80 MHz, jakost polja treba biti manja od 3 V/m.
a

140

141

VALO™ GRANDCS

POLYMERAČNÍ LAMPA
Popis produktu

Polymerační lampa VALO Grand je díky svému širokopásmovému spektru určena k polymerizaci všech světlem vytvrzovaných materiálů v rozsahu vlnových délek 385-515 nm podle normy
ISO 10650. Polymerační lampa VALO Grand používá dobíjecí baterie a nabíječku baterií Ultradent VALO. Polymerační lampa je navržena tak, aby se dala umístit do standardního držáku
zubní jednotky, nebo ji lze namontovat na zakázku pomocí držáku, který je součástí sady.

Součásti produktu VALO velká:
•
•
•
•
•
•

Fotografie

Aby nedošlo k poškození zařízení, NEVKLÁDEJTE prsty, nástroje ani jiné předměty do prostoru pro baterie polymerační lampy.
Aby nedošlo k poškození zařízení, NEPOKOUŠEJTE se čistit zlaté kontakty ani žádnou část prostoru pro baterie. V případě pochybností volejte zákaznický servis Ultradent.
Aby se zabránilo riziku křížové kontaminace, ochranné návleky jsou určeny pouze pro použití u jednoho pacienta.
Abyste snížili riziko koroze, ochranný návlek po použití odstraňte.
Abyste snížili riziko nedostatečného vytvrzení pryskyřice, nepoužívejte polymerační lampu, pokud je poškozená čočka.

Postupné pokynyPříprava

1) Před použitím polymerační lampy nabijte baterie (viz část Údržba baterií).
2) Před každým použitím nasaďte na polymerační lampu nový ochranný návlek a minimalizujte záhyby na čočce, abyste dosáhli co nejlepších výsledků.

Poznámka:

1 – Polymerační lampa VALO Grand4 – Nabíjecí baterie VALO Ultradent
1 – Nabíječka baterií VALO Ultradent s napájecím zdrojem medical grade 12VDC AC
1 – Balení vzorků ochranných návleků VALO
1 – Ochranný štítek pro polymerační lampu VALO Grand1 – Montážní držák na VALO s oboustrannými samolepkamiPřehled ovládání:

•
•
•
•
•

• Aby se zabránilo křížové kontaminaci a ulpívání dentálního kompozitního materiálu na povrchu čočky a těla zařízení, je třeba při každém použití polymerační lampy VALO použít
ochranný návlek schválený společností Ultradent. Ochranné návleky jsou určeny pro použití u jednoho pacienta.

Světelný štít VALO bezdrátová:Světelný štít VALO bezdrátová je oválný, může být otočen pro maximální použití a může být použit s průhledným ochranným pouzdrem.

PoužitíTlačítka a světla

LegendaTlačítko(a) napájení
(Horní nebo spodní)Světla časování
Slabá baterie – indikátorSvětlo režimu/stavu
Tlačítko časování / změny
režimu1) Každý režim napájení se používá pro vytvrzování dentálních materiálů pomocí foto-iniciátorů. Doporučené doby vytvrzování naleznete v rychlém průvodci režimy.
NOTE: Polymerační lampa je naprogramována, aby postupně přecházela z režimu Standardní výkon do režimu Vysoký výkon plus a následně do režimu Extra výkon. Chcete-li například
přepnout z režimu Standardní výkon do režimu Extra výkon, je nutné přepnout do režimu Vysoký výkon plus a poté do režimu Extra výkon.
2) Polymerační lampa vždy uloží naposledy použitý interval načasování a režim a navrátí se k nim, kdykoli dojde ke změně režimů nebo když jsou vytaženy baterie.

ProvozREŽIM TVRZENÍ: Standardní režim napájení

INTERVALY NAČASOVÁNÍ: 5, 10, 15, a 20 sekund.
• Při PRVNÍM zapnutí má polymerační lampa nastaven tento režim. Kontrolka režimu/stavu bude svítit zeleně a rozsvítí se čtyři zelené kontrolky časování, což indikuje režim Standardnívýkon.
• Pro změnu intervalů načasování krátce stiskněte tlačítko času/režimu.
• Polymeraci spustíte stiskem tlačítka napájení. Chcete-li přestat polymerovat, než doběhne časový interval, znovu stiskněte tlačítko napájení.

REŽIM TVRZENÍ: Režim Vysoký výkon plus

Zdroj osvětlení pro vytvrzování fotoaktivních dentálních výplňových materiálů a lepidel.

INTERVALY NAČASOVÁNÍ: 1, 2, 3, a 4 sekundy.
• V režimu Standardní výkon stiskněte a podržte tlačítko pro změnu času/režimu po dobu 2 sekund a uvolněte. Kontrolka režimu/stavu bude svítit oranžově a pak se rozsvítí a začnou
blikat čtyři zelené kontrolky časování, což indikuje režim Vysoký výkon.
• Pro změnu intervalů načasování krátce stiskněte tlačítko času/režimu.
• Polymeraci spustíte stiskem kteréhokoli tlačítka Power (Napájení). Chcete-li přestat polymerovat, než doběhne časový interval, znovu stiskněte kteerékoli tlačítko Power (Napájení).
• Pro návrat do režimu Standardní výkon stiskněte a podržte tlačítko pro změnu času/režimu po dobu 2 sekund, čímž přejdete do režimu Extra výkon. Pak tlačítko pusťte, podržte po
dobu 2 sekund a znovu pusťte. Kontrolka režimu/stavu bude svítit zeleně a rozsvítí se čtyři zelené kontrolky časování, což indikuje režim Standard.

Varování a bezpečnostní opatření

REŽIM TVRZENÍ: Režim Extra výkon

(Dole)(Nahoře)Před použitím kteréhokoli produktu si pečlivě přečtěte všechny pokyny a bezpečnostní listy a porozumějte jim.

Indikace použití / zamýšlený účelRiziková skupina 2
UPOZORNĚNÍ Výrobek vyzařuje UV světlo. Při expozici může dojít k podráždění očí nebo kůže. Použijte vhodné stínění.
UPOZORNĚNÍ Možné nebezpečné optické záření emitované tímto produktem. Nedívejte do lampy, která je v provozu. Může dojít k poškození očí.
• NEDÍVEJTE se přímo do světelného zdroje. Pacient, lékař a asistenti by měli při použití polymerační lampy vždy používat oranžově zbarvenou ochranu očí.
• Z důvodu zabránění nebezpečí úrazu elektrickým proudem nejsou povoleny žádné úpravy tohoto zařízení. Používejte pouze přiložený napájecí zdroj a zástrčkové adaptéry Ultradent
VALO. Pokud jsou tyto součásti poškozené, nepoužívejte je, zavolejte do zákaznického servisu Ultradent a objednejte si náhradní.
• Přenosné RF komunikační zařízení může zhoršit výkon, pokud je použito blíže než 30 cm od zařízení.
• Používejte pouze autorizované příslušenství, kabely a napájecí zdroje, abyste zabránili nesprávnému provozu, zvýšenému elektromagnetickému vyzařování nebo snížené
elektromagnetické odolnosti (viz část Elektromagnetické vyzařování).
• Aby se zabránilo riziku požáru elektrického proudu spojeného s manipulací s bateriemi:
⸰ NEMÍCHEJTE dobíjecí baterie s nedobíjecími bateriemi nebo jinými typy baterií.
⸰ NEPOKOUŠEJTE se nabíjet nedobíjecí baterie. Nabíjení nedobíjecích baterií může mít za následek zranění nebo poškození majetku.
⸰ NEPOUŽÍVEJTE autokláv a na baterii, kontakty baterie, nabíječku nebo síťový zdroj nestříkejte žádnou tekutinu. Pokud se na kontaktech nabíječky objeví koroze, zavolejte dozákaznického servisu Ultradent a objednejte si výměnu.
⸰ NENABÍJEJTE baterie v blízkosti hořlavých materiálů.
⸰ NENECHÁVEJTE nabíječku na klinické operační stanici.
• Abyste předešli riziku zranění, NEPOUŽÍVEJTE baterie, které jsou zkorodované (rezavé), promáčknuté, vydávají zápach nebo tekutiny, mají roztržený nebo chybějící obal nebo jsou jinak
poškozené. Zavolejte na zákaznický servis Ultradent a objednejte si nové baterie.
• Používejte pouze doporučené baterie. Jiné baterie mohou způsobit poruchy.
• Abyste předešli riziku tepelného podráždění nebo poranění, vyhněte se několika cyklům vytvrzování za sebou a nevystavujte měkké tkáně dutiny ústní těsné blízkosti světlu po dobu
delší než 10 sekund, a to v jakémkoli režimu. Pokud je vyžadována delší doba vytvrzování, použijte více vytvrzovacích cyklů s přestávkami mezi cykly nebo použijte produkt s duálním
vytvrzováním, abyste tak zabránili zahřívání měkkých tkání.
• Buďte opatrní při léčbě pacientů, kteří trpí nežádoucími fotobiologickými reakcemi nebo citlivostí, při léčbě pacientů, kteří podstupují chemoterapii, nebo při léčbě pacientů léčených
fotosenzibilizujícími léky
• Tato jednotka může být náchylná na silná magnetická nebo statická elektrická pole, která by mohla narušit programování. Pokud máte podezření, že k tomu došlo, odpojte na okamžik
přístroj ze zásuvky a poté jej znovu zapojte.
• Polymerační lampu NEUTÍREJTE žíravými nebo abrazivními čisticími prostředky, nevkládejte do autoklávu ani do žádné ultrazvukové lázně, dezinfekčního prostředku, čisticího roztoku
nebo kapaliny. Nedodržení přiložených pokynů pro postup může způsobit nefunkčnost polymerační lampy.
142

INTERVAL NAČASOVÁNÍ: Pouze 3 sekundy (Poznámka: Režim Extra výkon má na konci každého cyklu 2sekundové bezpečnostní zpoždění, aby bylo omezeno přehřívání během po sobějdoucího tvrzení. Na konci zpoždění se ozve zapípání značící, že je jednotka připravena pro nepřetržité použití.
• V režimu Standardní výkon stiskněte tlačítko pro změnu času/režimu po dobu 2 sekund, tlačítko pusťte, znovu stiskněte a držte po dobu 2 sekund a pusťte. Kontrolka režimu/stavu
bude svítit oranžově a blikat, pak se rozsvítí a začnou blikat tři zelené kontrolky časování, což indikuje režim Extra výkon.
• Polymeraci spustíte stiskem tlačítka napájení. Chcete-li přestat polymerovat, než doběhne časový interval, znovu stiskněte tlačítko napájení.
• Pro návrat do režimu Standardní výkon stiskněte a podržte tlačítko času/režimu po dobu 2 sekund a pak pusťte. Kontrolka režimu/stavu bude svítit zeleně a rozsvítí se zelené
kontrolky časování, což indikuje režim Standardní výkon.

Režim spánku: Polymerační lampa přejde do režimu spánku po 60 hodině nečinnosti. Režim spánku značí pomalu blikající světlo režimu/stavu. Polymerační lampu probudíte zvednutím
nebo dotykem jednotky. Lampa se automaticky navrátí do posledního použitého nastavení. Chcete-li prodloužit životnost baterií, polymerační lampy se nedotýkejte, když se nepoužívá.

Čištění1) Po každém použití vyhoďte použité ochranné návleky do běžného odpadu.
2) Viz část Zpracování.

Pokyny pro montáž držáku

1) Držák připevněte na rovný, nemastný povrch.
2) Povrch očistěte alkoholem.
3) Odlepte papír z lepicí pásky.
4) Umístěte držák, aby lampa při vytahování směřovalo vzhůru. Pevně zatlačte.

143

Rychlý průvodce režimy:Standardní výkon
1000 mW/cm2

Režim

Vysoký výkon plus
1600 mW/cm2

Extra výkon
3200 mW/cm2

Tlačítko napájení
LED režimu/načasování

Jak nabíjet baterie:1) Nabíječku a baterie umístěte na místo, kde nepřijdou do styku s kapalinami nebo s dezinfekčním sprejem.
2) Vložte baterie do nabíječky kladným (+) koncem směrem ke kontrolkám na nabíječce.
3) Zelené kontrolky indikují, že baterie jsou připraveny k použití.
4) Nabíjení baterií trvá 1-3 hodiny. Baterie nechte v nabíječce, dokud nebudou připraveny k použití.
POZNÁMKA: Pokud se červená kontrolka na nabíječce nerozsvítí zeleně, když se baterie nabíjejí déle než tři hodiny, může být baterie znehodnocená (poškozená) a nelze ji nabíjet.
Vyzkoušejte novou baterii nebo zavolejte do zákaznického servisu Ultradent a objednejte si novou sadu dobíjecích baterií.

Tlačítka časuMožnosti času

5s

10s

15s

20s

1s

2s

3s

Pro změnu časuPro přechod mezi časovými možnostmi stiskněte krátce tlačítko času.

Pro změnu režimu

Stiskněte tlačítko času po dobu 2 sekund. Polymerační lampa přejde do dalšího režimu.

LegendaSvítící LED

4s

Pouze 3s

Blikající LED

Rychlý průvodce tvrzení:Režim vytvrzováníVýkon* (mW)Zářivost* (mW/cm2)

Celková doba expozice (v sekundách)

Energie** (Jouly)

Standard

970

900

5/10/15/20

4.8J/9.7J/14.5J/19.4J

High Power Plus

1,620

1,500

1/2/3/4

1.6J/3.2J/4.8J/6.5J

Xtra

2,260

2,100

3

6,8 J

*Jmenovitý výkon ve vzdálenosti 2 mm od hrotu čočky k hornímu povrchu kompozitu.
**Hodnoty v oddíle celková energie na cyklus (jouly) jsou zaokrouhleny na nejbližší desetinu.

Rychlý průvodce varováními:
Varování úrovně napájení

Upozornění na teplotu

Servisní varování

LED VarováníVyměňte baterie

Nechte vychladnout

Zavolejte na zákaznické služby pro
informace o opravě

Zavolejte na zákaznické služby pro
informace o opravě

• Žádný zvuk
• Bliká každé 2 sekundy• Umožňuje provoz• Nepřetržité 3 zapípání
• Jedno rychlé zablikání• Zakazuje provoz• Slabá baterie: pomalé blikání• Vypněte: 3 pípnutí, blikání• Zakazuje provoz

1) Používejte ochranný návlek, abyste zabránili ulpívání dentálního kompozitu na povrchu čočky. V případě potřeby opatrně odstraňte nalepený kompozit pomocí plastového nebo
nerezového zubního nástroje. Nepoužívejte nástroje, které by poškodily čočku.
2) Měřiče světla se značně liší a jsou určeny pro konkrétní světlovodné koncovky a čočky. Ultradent doporučuje rutinně kontrolovat výstup v režimu Standard Power (Standardní výkon).
POZNÁMKA: skutečný číselný výstup bude zkreslený v důsledku nepřesnosti běžných měřičů světla a vlastního bloku LED, které je v polymerační lampě.
Údržba baterie
Nabíjení a výměna baterií
Polymerační lampa je dodávána se 4 dobíjecími lithium-železo-fosfátovými bateriemi. Ultradent doporučuje používat pouze dobíjecí baterie VALO.

• 3 pípnutí• Pomalé blikání• Zakazuje provoz

Jak vyměnit/vložit baterie:
1) Odstraňte zadní kryt otočením proti směru hodinových ručiček o jednu čtvrtinu otáčky.
2) Vyjměte baterie.
3) Nejprve vložte kladnou (+) stranu nových baterií.
4) Zadní kryt nasaďte zpět tak, že jej vyrovnáte a jemně zatlačíte při otáčení ve směru hodinových ručiček. Po úplném nasazení víčko zacvakne.
5) Přístroj je připraven k použití.
Slabé baterie: Kontrolka vytvrzování signalizuje uživateli, že je čas vyměnit baterie, když kontrolka indikující slabé baterie bliká červeně. Pokud dojde k příliš nízkému nabití baterií, ozve
se zvukové varování se třemi pípnutími. Polymerační lampa neumožní další provoz, dokud nebudou baterie dobity nebo dokud nebudou vloženy baterie nové. (Viz Rychlý průvodce
varováním).Doba nabíjení a životnost baterie: Životnost plně nabité baterie v polymerační lampě závisí na režimu/časovém intervalu, typu baterie, množství použití a účinnosti LED.
• Doporučený interval dobíjení: V případě potřeby, nebo když se rozsvítí indikátor vybité baterie. Očekávaná životnost závisí na používání.
• Extra baterie: Součástí dodávky polymerační lampy jsou 4 dobíjecí baterie.
Oprava výrobcem
1) Opravy smí provádět pouze autorizovaný servisní personál. Ultradent poskytuje servisním pracovníkům dokumentaci k provádění oprav.
2) Při odesílání přístrojů k opravě, k servisu nebo ke kalibraci vždy vyjměte z polymerační lampy baterie a nabíječky. Baterie, nabíječku, adaptér a polymerační lampu zabalte zvlášť dovratné krabice.
3) Baterie odesílejte v souladu s místními předpisy.
ZárukaSpolečnost Ultradent Products, Inc. („Ultradent“) zaručuje, že tento výrobek bude po dobu 5 let od data zakoupení, pokud bude provozován v souladu s návodem k obsluze přiloženým kvýrobku, (i) ve všech podstatných ohledech odpovídat specifikacím uvedeným v dokumentaci společnosti Ultradent přiložené k výrobku; a (ii) bude bez vad materiálu a zpracování.
Tato omezená záruka je nepřenosná a vztahuje se pouze na původního kupujícího a nevztahuje se na další majitele výrobku. Tato omezená záruka se nevztahuje na žádné další součásti
příslušenství, jako jsou mimo jiné baterie, nabíječky, adaptéry nebo adaptivní objektivy. Tato omezená záruka je neplatná, pokud výrobek selže nebo je poškozen v důsledku nedbalosti,zneužití, nesprávného použití, nehody, úpravy, zásahu, změny nebo nedodržení příslušných pokynů k použití. Jako příklad, tato záruka se nevztahuje na výrobek, který byl upuštěn apoškozen. Aby bylo možné uplatnit tuto omezenou záruku, musí být společnosti Ultradent spolu s vadným výrobkem předložen doklad o koupi (např. prodejní doklad nebo podobná
dokumentace).
Vadný výrobek, který splňuje zde uvedené záruční podmínky, bude podle uvážení společnosti Ultradent buď opraven, nebo vyměněn. Odpovědnost společnosti Ultradent za výrobek v
žádném případě nepřesáhne kupní cenu zaplacenou kupujícím. Společnost Ultradent za žádných okolností neodpovídá za jakékoli nepřímé, náhodné, předvídané, nepředvídané, zvláštní
nebo následné škody vzniklé v důsledku používání tohoto výrobku nebo v souvislosti s ním.

Zpracování

Tento výrobek je určen k použití s ochranným návlekem. Údržba a běžné čištění viz výše.

Skladování a likvidace

Pokud uskladňujete vytvrzovací světlo po dobu delší než 2 týdny nebo jej balíte pro cestování, vždy vyjměte baterie. Pokud jsou baterie ponechány v přístroji po delší dobu bez dobíjení,mohou se stát nefunkčními nebo je nemusí být možné dobít. Neskladujte baterie při teplotách nad 40 °C (104 °F) nebo na přímém slunečním světle.
Podmínky skladování a přepravy polymerační lampy:• Teplota: +10 °C až +40 °C (+50 °F až +104 °F)• Relativní vlhkost: 10 % až 95 %• Okolní tlak: 500 hPa až 1060 hPaÚdržbaObecné čištění polymerační lampy
Po každém použití navlhčete gázu nebo měkký hadřík schváleným dezinfekčním prostředkem na povrchy a otřete povrch a čočku. Neschválené čisticí prostředky mohou způsobit
poškození polymerační lampy.
PŘÍPUSTNÉ ČISTICÍ PROSTŘEDKY:• 70% Isopropylalkohol• 70% Etanol

Obecné čištění světelného štítu:
Očistěte bezdrátový světelný štít VALO pomocí jakéhokoli povrchového dezinfekčního prostředku. NEPOUŽÍVEJTE autokláv.
OpravaÚdržba prováděná uživatelem144Při likvidaci elektronického odpadu (tj. polymeračních lamp, nabíječek, baterií a napájecích zdrojů) postupujte podle místních směrnic pro likvidaci odpadu a recyklaci.

Technická hlediskaPříslušenstvíPoložka

Informace o CESvětelný štít VALO bezdrátová:

145

Položka

Informace o CE

Technické informace/údaje

Pásma vlnových délek efektivního tvrzení kompozitu:

Nabíječka VALOEFEKTIVNÍ KOMPOZITNÍ VYTVRZOVACÍ VLNOVÁ PÁSMA
(Použitelný rozsah vlnových délek: 385-515 nm)
SPEKTRÁLNÍ ZÁŘIVÝ VÝKON (mW/nm)

Baterie VALO

Efektivní nižší
pásmo vlnových délek
395-415 nm

Efektivní pásmo horní
vlnové délky
440-480 nm

XtraHigh Power (Vysoký výkon)Standard

WAVELENGTH (nm)Atribut

Informace/specifikaceČočkaPrůměr 11,7 mm

Rozsah vlnových
délek• Použitelný rozsah vlnových délek: 385 – 515 nm• Maximální vlnové délky: 395 – 415 nm a 440 – 480 nm
Graf porovnávající jmenovité vyzařováníPřístroj pro měření

Tabulka svítivosti

146

Spektrální analyzátor ‡ Gigahertz
Exitance

Total Power

Apertura

15 mm

15 mm

Režim Standardnívýkon (±10%)900 mW/cm²

970 mW

Vysoký výkon plus(±10%)

1500 mW/cm²

1615 mW

Extra výkon (±10%)2100 mW/cm²

Exitance záření se bude lišit v závislosti namožnostech přístroje, metodě měření aumístění lampy.
‡ Exitance záření odpovídá normě ISO
10650 při měření spektrálním analyzátorem
Gigahertz-Optik.

2260 mW

Polymerační lampa
VALO velká

Charakteristika: IEC 60601-1 (Bezpečnost), IEC
60601-1-2 (EMC)

Hmotnost:• S bateriemi: 6 oz. (190 gramů)• Bez baterií: 5 oz. (150 gramů)Rozměry: (8 x 1,28 x 1,06) palce, (203 x 32,5 x 27) mm

Napájecí zdroj
nabíječkyVýstup - 12VDC při 500mAPříkon – 100 V až 240 V (střídavý proud)
Ultradent P/N 5930 VALO Napájecí zdroj s
mezinárodními konektory

Charakteristika: IEC 60601-1 (Bezpečnost)
Délka kabelu - 6 stop (1,8 metru)
Napájecí zdroj VALO nabíječky je zdroj napájení lékařské třídy II a zajišťuje izolaci odhlavního přívodu

Nabíječka VALOInteligentní nabíječka baterií VALO 3,6 VDC (lithium-železitý fosfát):• Automatické vypnutí při plném nabití• Automatická detekce vadných baterií• Ochrana: tepelná, přebíjení, zkrat, zpětná polarita⸰ Červená LED - nabíjení
⸰ Zelená LED - vybitá nebo plně nabitá
⸰ LED nesvítí - zkrat• Doba nabíjení: 1 až 3 hodiny
Charakteristika: CE, WEEE
147

Atribut

Informace/specifikace

Baterie VALO

Nabíjecí: Bezpečný lithium-železitý fosfát (LiFePO4) RCR123A• Pracovní napětí: 3,2VDC
Charakteristika: UL, CE, RoHS, WEEE

Provozní podmínky

Teplota: +10 °C až +32 °C (+50 °F až +90 °F)
Relativní vlhkost: 10 % až 95 %Okolní tlak: 700 hPa až 1060 hPa

Pracovní cyklusRůzné informace
Poučení a prohlášení výrobce ohledně elektromagnetických emisí
Vytvrzovací světlo je určeno pro použití v elektromagnetickém prostředí specifikovaném níže. Zákazník nebo uživatel by měli zajistit, aby světlo bylo používán v takovém
prostředí.

Polymerační lampa je navržena pro krátkodobý provoz. Používejte při maximální teplotě okolí 32 °C po dobu maximálně 1 minuty po soběnásledujícího tvrzení a poté na dobu 30 minut VYPNĚTE (doba pro ochlazení lampy).

Odstraňování problémů
Pokud níže uvedená řešení nevyřeší problém, s kterým se potýkáte, obraťte se na společnost Ultradent na čísle 800.552.5512. Mimo Spojené státy zavolejte svému
distributorovi značky Ultradent nebo dentálnímu dealerovi.
Problém

Možná řešeníSvětlo se nerozsvítí

1) Stiskněte tlačítko času/režimu nebo napájecí tlačítko, kterým lampu probudíte z úsporného režimu.
2) Stav baterie vám prozradí červený ukazatel slabé baterie.
3) Zkontrolujte, zda jsou do přístroje správně vloženy nové baterie.
4) Pokud blikají červené a žluté výstražné LED diody, znamená to, že polymerační lampa dosáhla svého interního
bezpečnostního limitu teploty. Nechte vytvrzovací světlo vychladnout po dobu 10 minut nebo použijte chladný vlhký
ručník, aby se jednotka rychle ochladila.
5) Pokud červená výstražná dioda LED bliká a nepřetržitě pípne, obraťte se na zákaznický servis Ultradent.

Světlo nezůstane rozsvícené popožadovanou dobu

1) Zkontrolujte správné nastavení času světel režimu a časování.
2) Stav nabití baterie prozradí ukazatel slabé baterie.
3) Zkontrolujte, zda jsou do jednotky správně vloženy nové baterie.

Světlo nepolymeruje pryskyřici, jakby mělo1) Zkontrolujte, zda se na čočce nenachází vytvrzená pryskyřice/kompozity.
2) Použijte vhodné oranžové ochranné brýle před UV světlem a ověřte, zda fungují LED světla.
3) Pomocí expozimetru zkontrolujte hladinu výkonu. Pokud používáte expozimetr, společnost Ultradent doporučuje kontrolu
polymerační lampy v režimu Standardní výkon.
POZNÁMKA: Skutečný číselný výsledek bude zkreslený kvůli nepřesnosti běžných expozimetrů a zakázkového LED osvětlení,které polymerační lampa využívá. Expozimetry se výrazně odlišují a jsou navrženy pro konkrétní špičky a čočky vodičů světla.
4) Zkontrolujte datum expirace používané pryskyřice.
5) Přesvědčte se, zda používáte vhodnou techniku podle doporučení výrobce.

Baterie se nenabíjí

1) Ujistěte se, že jsou baterie vloženy do nabíječky ve správné orientaci a nechte baterie nabíjet po dobu 1 až 3 hodin.
2) Pokud se červená kontrolka na nabíječce nezmění na zelenou, zavolejte na zákaznický servis Ultradent a objednejte si
náhradní baterie a/nebo nabíječku.
3) Pokud není na nabíječce viditelná ani zelená, ani červená kontrolka, zavolejte na zákaznický servis Ultradent a objednejte
si nebo vyměňte nabíječku a/nebo síťový adaptér.

Nabíječka nenabíjí baterie

1) Ujistěte se, že je nabíječka připojena a síťový adaptér je zapojen do fungující elektrické zásuvky.
2) Pokud na nabíječce není viditelná zelená nebo červená kontrolka, zavolejte na zákaznický servis Ultradent pro objednávku
nové nabíječky a/nebo síťového adaptéru.

Nelze změnit režim nebo časové
intervaly

1) Podržte tlačítka čas/režim a tlačítko napájení, dokud řada pípnutí neoznačí, že je polymerační světlo odemčeno.

Zkouška emisíDodržování předpisů

Elektromagnetické prostředí - poučení

RF emiseCISPR 11

Skupina 1

Polymerační lampa používá baterie a nemá na ni vliv EMI, RF nebo potlačené přepětí.

RF emiseCISPR 11Třída B

Vytvrzovací světlo používá elektrickou a elektromagnetickou energii pouze pro své vnitřní funkce.
Proto jsou veškeré RF emise velmi nízké a není pravděpodobné, že by způsobovaly rušení v blízkých
elektronických zařízeních.

Harmonické emise
IEC 61000-3-2

Nevztahuje se

Testování emisí harmonických a kolísání napětí se nevztahuje na polymerační lampu, protože jenapájena z baterie.

Kolísání napětí / emise blikáním Nevztahuje se
IEC 61000-3-3

Polymerační lampa je vhodná pro použití ve všech zařízeních, včetně domácích zařízení a zařízení
přímo napojených na veřejnou nízkonapěťovou síť, která zásobuje budovy pro domácí použití.

Poučení a prohlášení výrobce ohledně elektromagnetické imunity
Vytvrzovací světlo je určeno pro použití v elektromagnetickém prostředí specifikovaném níže. Zákazník nebo uživatel by měli zajistit, aby světlo bylo používán v
takovém prostředí.
Test IMUNITY

Testovací úroveň IEC 60601

Úroveň shody

Elektromagnetické prostředí - poučení

Elektrostatický výboj (ESD)± 8 kV kontakt± 15 kV vzduch± 8 kV kontakt± 15 kV vzduch

Fyzické prostředí by mělo být omezeno na následující:
1) Kód IP: IP20
2) Neponořujte do kapaliny.
3) Nepoužívejte v blízkosti hořlavého plynu.
Jednotka je non-APG a non-AP.
4) Rozsah vlhkosti při skladování 10 % až 95 %5) Rozsah skladovacích teplot: 10 °C až 40 °C

IEC 61000-4-2

Elektrické rychlé přechodnénapětí/prasknutí
IEC 61000-4-4± 2 kV pro napájecí vedení± 2 kV pro napájecí vedení± 1 kV pro vstupní/výstupní vedení Poznámka 1: Polymerační lampa
nemá žádné I/0 porty

Náraz
IEC 61000-4-5± 1 kV vedení do vedení± 2 kV vedení do země± 1 kV vedení do vedení± 2 kV vedení do zeměVzhledem k tomu, že polymerační lampa je napájena
z baterie, není vystavena elektrickým přechodným
proudům, přepětím, poklesům napětí, zkratům,přerušením nebo změnám výkonu hlavního vedení.

Napětí, poklesy, zkraty, přerušení
a odchylky na vstupních vedeních
napájení<5 % U(> 95% ponoření do U po dobu
0,5 cyklu)<5 % U(> 95% ponoření do U po dobu
0,5 cyklu)

Nabíječka baterií může podléhat výše uvedeným
pravidlům, ale na provoz polymerační lampy VALO
bezdrátová nemá zásadní vliv.

IEC 61000-4-11

40 % U(60% ponoření do U po dobu
5 cyklů)40 % U(60% ponoření do U po dobu
5 cyklů)70 % U(30% ponoření do U po dobu
25 cyklů)70 % U(30% ponoření do U po dobu
25 cyklů)Pokud napětí baterie polymerační lampy klesne na
4VDC, jednotka nepovolí provoz. Polymerační lampa
VALO bezdrátová se vypne. Když jsou vloženy novébaterie a dojde k obnovení napětí, lampa VALO serestartuje a vrátí se do stejného stavu, v jakém bylapřed výpadkem napájení. V případě výpadku napájení
se polymerační lampa samočinně obnoví.

<5 % U(> 95% ponoření do U po
dobu 5 s)<5 % U(> 95% ponoření do U po
dobu 5 s)Poznámka 2: Sebeobnovuje se

30 A/m30 A/mSíťový kmitočet(50/60 Hz) magnetické pole
IEC 61000-4-8

Polymerační lampa je napájena z baterie a neníschopna se připojit k hlavnímu vedení.

Magnetická pole síťového kmitočtu by měla být naúrovních charakteristických pro typické umístění
v typickém, obytném, domácím zdravotnickém,
komerčním, nemocničním nebo vojenském prostředí.

Poznámka 1: Polymerační lampa není vybavena žádnými porty ani žádnými přístupovými I/O linkami.

148

149

Poučení a prohlášení výrobce ohledně elektromagnetické imunity systému nepodporujících život

Pokyny a prohlášení výrobce pro doporučené odstupy mezi přenosným a mobilním RF komunikačním zařízením a světlem polymerační lampou VALO bezdrátová.

Vytvrzovací světlo je určeno pro použití v elektromagnetickém prostředí specifikovaném níže. Zákazník nebo uživatel by měli zajistit, aby světlo bylo používán v
takovém prostředí.

Polymerační lampa je určena pro použití v elektromagnetickém prostředí, ve kterém jsou řízeny vyzařované vysokofrekvenční rušení. Uživatel polymerační lampy můžepomoci zabránit elektromagnetickému rušení udržováním minimální vzdálenosti mezi přenosným a mobilním RF komunikačním zařízením (vysílače) a polymeračnílampou, jak je doporučeno níže, podle maximálního výstupního výkonu komunikačního zařízení.

Test IMUNITY

Testovací úroveň IEC
60601

Úroveň shody

Vedení RF

3 Vrms3 VrmsIEC 61000-4-6

150 kHz až 80 MHz

150 kHz až 80 MHzVyzařovaná RF

3 V/m

3 V/m

Elektromagnetické prostředí - poučeníPřenosné a mobilní RF komunikační zařízení by neměla být používána blíže žádné části
polymerační lampy, včetně kabelů, než je doporučená vzdálenost, která se vypočítá z
rovnice platné pro frekvenci vysílače.
Doporučená vzdálenost:

80 MHz až 800 MHz

800 MHz až 2,5 GHz
P je maximální výstupní výkon vysílače ve wattech (W) podle výrobce vysílače a d jedoporučená vzdálenost v metrech (m).
Intenzita pole z pevných vysílačů RF, jak je stanoveno elektromagnetickým průzkumem
místaa, by měla být menší než úroveň shody v každém kmitočtu rangeb.

IEC 61000-4-3

80 MHz až 2,5 GHz

80 MHz až 2,5 GHz

V blízkosti zařízení označených následujícím symbolem může dojít k rušení:Oddělovací vzdálenost podle frekvence vysílače
(metrů)Jmenovitý maximální výstupní výkon
vysílače(P ve wattech)

150 kHz – 80 MHz

80 MHz – 800 MHz

800 MHz – 2,5 GHz

0,01

0,12 metrů

0,035 metrů

0,07 metrů

0,10,37 metrů

0,11 metrů

0,22 metrů

1

1,7 metrů

0,35 metrů

0,7 metrů

10

3,7 metrů

1,11 metrů

2,22 metrů

100

11,7 metrů

3,5 metrů

7,0 metrů

Polymerační lampa byla testována podle normy IEC 60601-1-2:2014 a prošla v rámci vyzařovaných sil pole 10 V/m v rozsahu 80 V MHz až 2,5 GHz. Hodnota 3Vrms
odpovídá V1 a hodnota 10 V / m odpovídá E1 ve výše uvedených vzorcích.
Pro vysílače s jmenovitým maximálním výstupním výkonem, které nejsou uvedeny výše, může být doporučená vzdálenost d v metrech (m) odhadnuta pomocí rovniceplatné pro kmitočet vysílače, kde P je maximální výstupní výkon vysílače ve wattech ( W) podle výrobce vysílače.
POZNÁMKA 1 Při 80 MHz a 800 MHz platí vzdálenost pro vyšší frekvenční rozsah.
POZNÁMKA 2 Tyto pokyny nemusí platit ve všech situacích. Elektromagnetické šíření je ovlivněno absorpcí a odrazem od konstrukcí, objektů a osob.

POZNÁMKA 1 Při 80 MHz a 800 MHz platí vyšší frekvenční rozsah.
POZNÁMKA 2 Tyto pokyny nemusí platit ve všech situacích. Elektromagnetické šíření je ovlivněno absorpcí a odrazem od konstrukcí, objektů a osob.
Intenzity pole z pevných vysílačů, jako jsou základní stanice pro rádiové (mobilní/bezdrátové) telefony a pozemní mobilní rádia, amatérské rádio, rozhlasové vysílání
AM a FM a televizní vysílání, nelze teoreticky s přesností předpovědět. Pro posouzení elektromagnetického prostředí v důsledku pevných RF vysílačů by měl být zvážen
elektromagnetický průzkum místa. Pokud naměřená intenzita pole v místě, kde se používá polymerační lampa, překračuje příslušnou úroveň shody RF, je třeba
pozorovat polymerační lampu VALO bezdrátová, abyste ověřili normální provoz. Pokud je pozorován abnormální výkon, mohou být nutná další opatření, jako je změna
orientace nebo přemístění olymerační lampy.
b
Ve frekvenčním pásmu 150 kHz až 80 MHz by měly být intenzity pole nižší než 3 V/m.
a

150

151

VALO™ GRAND

ET

Curing Light

Toote kirjeldus

Oma lairibaspektriga on VALO Grand kõvestusvalgusti loodud polümeriseerima kõiki valguskõvastuvaid tooteid lainepikkuste vahemikus 385–515nm ISO 10650 järgi. VALO Grandkõvestusvalgustis kasutatakse Ultradent VALO laetavaid akusid ja akulaadijat. Kõvestusvalgusti on ette nähtud seisma tavalises hambaraviseadme hoidjas või selle saab kohandatultpaigaldada, kasutades komplektiga kaasasolevat konsooli.

VALO võimsa lambi osad:
•
•
•
•
•
•

1 – VALO Grand kõvestusvalgusti
4 – Ultradent VALO laetavad akud
1 – Ultradent VALO akulaadija meditsiinilise 12 V alalisvoolu vahelduvvooluadapteriga
1 – VALO Barrier Sleeve (kaitseümbris) näidispakk
1 – VALO Grand valguse kaitsekilp
1 – Kõvestusvalgusti paigaldusklamber kahepoolse teibiga

Juhtnuppude ülevaade:Sammsammulised juhised
Ettevalmistamine

1) Laadige akusid enne kõvestuslambi kasutamist (vt jaotist Aku hooldus).
2) Enne iga kasutuskorda asetage kõvestusvalgustile uus kaitseümbris, minimeerige kortse läätse ümber parima tulemuse saavutamiseks.
Märkus:• Ristsaastumise vältimiseks ja hamba komposiitmaterjali kleepumise vältimiseks läätse ja varre korpuse pinnale tuleb VALO kõvestusvalgusti kasutuskorral kasutada Ultradent
heakskiidetud kaitseümbrist. Kaitseümbrised on mõeldud kasutamiseks ühele patsiendile.
VALO juhtmeta lambi kaitse on ovaalne, seda saab maksimaalseks kasutamiseks pöörata ja läbipaistva kaitsmega kasutada.

Kasutamine

Nupud ja tuled

LegendToitenupp (toitenupud)(ülemised või alumised)
Ajastamistuled
Aku tühjenemine – indikaatorRežiimi/oleku tuli
Ajastuse/režiimi muutmise
nupp(Ülemine osa)

Lugege kõigi kirjeldatud toodete puhul enne kasutamist hoolikalt juhiseid ja ohutuskaardil olevat teavet ning veenduge, et saaksite kõigest aru.

Kasutusnäidustus/sihtotstarve

Valgusallikas fotoaktiveeritavate restauratiivsete hambaravimaterjalide ja -kleepainete kõvastamiseks.

Hoiatused ja ettevaatusabinõud
Risk Group 2
ETTEVAATUST! Toode väljastab UV-kiirgust. Kiirgusega kokkupuude võib põhjustada silmade või nahaärritust. Kasutage sobivaid kaitsevahendeid.
ETTEVAATUST! Toode väljutab potentsiaalselt ohtlikku optilist kiirgust. Ärge vaadake otse töötavasse lampi. See võib teie silmi kahjustada.
• ÄRGE vaadake otse valgusväljundisse. Patsient, arst ja abilised peaksid kõvestusvalguse kasutamisel alati kandma merevaiguvärvi UV-silmakaitset.
• Elektrilöögiohu vältimiseks ei ole seda seadet lubatud muuta. Kasutage ainult kaasasolevat Ultradent VALO toiteallikat ja pistikuadaptereid. Kui need komponendid on kahjustatud,ärge kasutage seadet ja helistage Ultradent klienditeenindusse, et tellida asendus.
• Kaasaskantavad raadiosageduslikud sideseadmed võivad jõudlust halvendada, kui neid kasutatakse lähemal kui 30 cm (12 tolli).
• Kasutage ainult volitatud tarvikuid, kaableid ja toiteallikaid, et vältida ebaõiget kasutamist, suurenenud elektromagnetkiirgust või vähenenud elektromagnetilist häireid (vt jaotist
Elektromagnetkiirgus).
• Akude käsitsemisega seotud elektritulekahju vältimiseks tehke järgmist.
⸰ ÄRGE segage laetavaid akusid mittelaetavate patareidega või muud tüüpi akudega.
⸰ ÄRGE proovige laadida mittelaetavaid akusid. Mittelaetavate akude laadimine võib põhjustada vigastusi või varalist kahju.
⸰ ÄRGE autoklaavige ega piserdage akusid, aku kontakte, laadijat ega vahelduvvoolu adapterit ühegi vedelikuga. Kui akulaadija kontaktidele ilmub korrosioon, helistage laadija
asenduse tellimiseks Ultradent klienditeenindusse.
⸰ ÄRGE laadige akusid süttivate materjalide läheduses.
⸰ ÄRGE hoidke laadijat kliinilises kasutuses.
• Vigastusohu vältimiseks ÄRGE kasutage akusid, mis on roostetanud (roostesed), mõlkis, lõhna või vedelikke eritavad, millel on rebenenud või puuduv ümbris või mis on muul viisil
kahjustatud. Asenduspatareide tellimiseks helistage Ultradent klienditeenindusse.
• Kasutage ainult soovitatud patareisid. Alternatiivsed akud võivad põhjustada talitlushäireid.
• Vältimaks termilise ärrituse või vigastuste ohtu, vältige vastastikku kõvenemise tsükleid ja ärge hoidke suu pehmeid kudesid üheski režiimis üle 10 sekundi vahetus läheduses. Kui on
vaja pikemat kõvenemisaega, kasutage mitut kõvendustsüklit koos puhkeperioodidega tsüklite vahel või kasutage pehmete kudede kuumutamise vältimiseks topeltkõvastuvat toodet.
• Olge ettevaatlik, kui ravite patsiente, kes kannatavad fotobioloogiliste kõrvaltoimete või tundlikkuse all, keemiaravi saavate patsientide või fotosensibiliseerivate ravimitega ravitavatepatsientide ravimisel.
• See seade võib olla vastuvõtlik tugevatele magnet- või staatilistele elektriväljadele, mis võivad programmeeringut häirida. Kui kahtlustate, et see on juhtunud, eemaldage seadehetkeks vooluvõrgust ja seejärel ühendage see uuesti pistikupessa.
• ÄRGE pühkige kõvestusvalgustit söövitavate või abrasiivsete puhastusvahenditega, ÄRGE autoklaavige ega kastke ultrahelivanni, desinfektsioonivahendisse, puhastuslahustesse egavedelikku. Kaasasolevate töötlemisjuhiste eiramine võib muuta kõvendusvalgusti töövõimetuks.
• Seadme kahjustamise vältimiseks ÄRGE sisestage sõrmi, instrumente ega muid esemeid kuumutuslambi patareipesasse.
152

Seadme kahjustamise vältimiseks ÄRGE püüdke puhastada kuldseid kontakte ega ühtegi patareipesa osa. Kui teil on arusaamatusi, helistage Ultradent klienditeenindusse.
Ristsaastumise ohu vältimiseks on kaitseümbrised ühe patsiendi jaoks kasutatavad.
Korrosiooniohu vähendamiseks eemaldage pärast kasutamist kaitseümbris.
Kui lääts on kahjustatud, ärge kasutage kõvendusvalgustit, et vähendada vaikude alakõvastumist.

VALO juhtmeta lambi kaitse:

Foto(Alumine osa)

•
•
•
•

1) Igat võimsuse režiimi kasutatakse fotoinitsiaatoritega hambaravimaterjalide kõvastamiseks. Vt soovitatavaid kõvastamisaegu režiimide kiirjuhendist.
MÄRKUS: Kõvastamislamp on programmeeritud liikuma järjest standardse võimsusega režiimist võimsasse erirežiimi ja sealt edasi eriti võimsasse režiimi. Näiteks standardsevõimsusega režiimist eriti võimsasse režiimi liikumiseks tuleb esmalt liikuda võimsasse erirežiimi ja sealt edasi eriti võimsasse režiimi.
2) Kõvastamislamp talletab viimati kasutatud ajastusintervalli ja režiimi ning valib selle vaikimisi iga kord, kui režiime muudetakse või kui akud eemaldatakse.

KasutamineKÕVASTAMISREŽIIM: Standardse võimsusega režiimAJASTUSINTERVALLID: 5, 10, 15, ja 20 sekundit.
• Kõvastamislamp valib vaikimisi selle režiimi ESMAKORDSE sisselülitamise korral. Režiimi/oleku tuli on roheline ja neli rohelist ajastustuld põlevad, tähistades standardse võimsusegarežiimi.
• Ajastusintervallide muutmiseks vajutage kiiresti aja/režiimi nuppu.
• Vajutage kõvastamiseks toitenuppu. Kõvastamise peatamiseks enne ajastusintervalli lõppu vajutage uuesti toitenuppu.

KÕVASTAMISREŽIIM: Võimas erirežiimAJASTUSINTERVALLID: 1, 2, 3, ja 4 sekundit.
• Hoidke standardse võimsusega režiimis 2 sekundit all aja/režiimi muutmise nuppu ja vabastage see. Režiimi/oleku tuli on oranž ja neli rohelist ajastustuld põlevad ning vilguvad,
tähistades võimsat režiimi.
• Ajastusintervallide muutmiseks vajutage kiiresti aja/režiimi nuppu.
• Vajutage kõvastamiseks üht toitenuppu. Kõvastamise peatamiseks enne ajastusintervalli lõppu vajutage uuesti üht toitenuppu.
• Hoidke standardse võimsusega režiimi naasmiseks 2 sekundit all aja/režiimi muutmise nuppu ning vabastage see, et liikuda eriti võimsasse režiimi. Hoidke nuppu uuesti 2 sekundit
all ja vabastage see. Režiimi/oleku tuli on roheline ja neli rohelist ajastustuld põlevad, tähistades standardse võimsusega režiimi.

KÕVASTAMISREŽIIM: Eriti võimas režiimAJASTUSINTERVALL: ainult 3 sekundit (Märkus. Eriti võimsal režiimil on iga kõvastamistsükli lõpus 2-sekundiline ohutusviivitus, et piirata järjestikuse kõvastamise korral kuumenemist.
Viivituse lõppemisest ja seadme jätkuvast kasutusvalmidusest annab märku piiksumine).
• Vajutage standardse võimsusega režiimis 2 sekundit aja/režiimi muutmise nuppu, vabastage see ning hoidke seda uuesti 2 sekundit all ja seejärel vabastage see. Režiimi/oleku tuli
on oranž ja vilgub ning kolm rohelist ajastustuld põlevad ja vilguvad, tähistades eriti võimsat režiimi.
• Vajutage kõvastamiseks toitenuppu. Kõvastamise peatamiseks enne ajastusintervalli lõppu vajutage uuesti toitenuppu.
• Standardse võimsusega režiimi naasmiseks hoidke 2 sekundit all aja/režiimi nuppu ning vabastage see. Režiimi/oleku tuli on roheline ja rohelised ajastustuled põlevad, tähistades
standardse võimsusega režiimi.

Unerežiim: kõvastamislamp läheb unerežiimi, kui seda pole 60 sekundit kasutatud, ja sellele viitab režiimi/oleku tule aeglane villkumine. Seadme kättevõtmine või puudutamine äratabkõvastamislambi üles, misjärel naaseb lamp automaatselt viimasele kasutatud sättele. Aku kasutusea maksimeerimiseks ärge puutuge kõvastamislampi, kui te seda parajasti ei kasuta.

Puhastamine

1) Visake kasutatud kaitsmed pärast iga patsienti tavajäätmete hulka.
2) Vt jaotist Töötlemine

Klambri paigaldamise juhised1) Klamber tuleb paigaldada tasasele õlivabale pinnale.
2) Puhastage pinda piiritusega.
3) Tõmmake klambri kleepribalt kate maha.
4) Asetage klamber nii, et kõvastamislamp liigub eemaldamise korral üles. Suruge klamber kindlalt paigale.

153

Režiimide kiirjuhend:Standardse võimsusega režiim
1000 mW/cm2

RežiimVõimas erirežiim
1600 mW/cm2

Eriti võimas režiim
3200 mW/cm2

Toitenupp

roostevabast terasest hambaraviinstrumenti. Ärge kasutage objektiivi kahjustavaid tööriistu.
2)
Valgusmõõturid on väga erinevad ja on mõeldud konkreetsete valgusjuhiotste ja läätsede jaoks. Ultradent soovitab väljundit regulaarselt kontrollida Standardvõimsuse režiimis. MÄRKUS: tegelik numbriline väljund on moonutatud tavaliste valgusmõõturite ebatäpsuse ja kuumutusvalguses kohandatud LED-paketi tõttu.
Aku hooldus
Akude laadimine ja vahetamineKõvestusvalgustiga on kaasas 4 taaslaetavat liitiumraudfosfaatakut. Ultradent soovitab kasutada ainult VALO laetavaid akusid.
Kuidas akusid laadida:
1)Asetage laadija ja akud kohta, kus need ei puutu kokku vedelike ega desinfektsioonivahendiga.
2)
Sisestage akud laadijasse nii, et positiivne (+) ots on suunatud laadija märgutulede poole.
Rohelised tuled näitavad, et patareid on kasutamiseks valmis.
3)
4)
Akude laadimine võtab aega 1-3 tundi. Jätke akud laadijasse kuni on kasutusvalmis.
MÄRKUS: Kui laadija punane tuli ei muutu roheliseks, peale kolme tundi laadimist, võib aku olla kahjustunud ja seda ei saa laadida. Proovige uut akut või helistage Ultradentklienditeenindusse, et tellida uus laetavate akude komplekt.

Režiimi/ajastuse LED-tuled

Aja nupud

Aja valikud

5s

10s

15s

20s

1s

2s

3s

4s

Aja muutmine

Vajutage aja nuppu ja vabastage see kiiresti, et liikuda läbi aja valikute.

Režiimi muutmine

Hoidke aja nuppu 2 sekundit all ja vabastage see. Kõvastamislamp liigub järgmisse režiimi.

LegendPõlevad LED-tuled

Ainult 3s

Vilkuvad LED-tuled

Tühjenevad akud: Kui kõvastusvalgusti märgutuli vilgub punaselt, annab see kasutajale märku, et on aeg akusid vahetada. Kui aku laetuse tase langeb liiga madalaks, kostab 3-piiksuline
hoiatusheli ja kõvestusvalgusti ei võimalda edasist tööd enne akude laadimist või uute patareide sisestamist. (Vt Kiirhoiatusjuhendit)Kõvastamise kiirjuhend:Kõvenemise režiim

Võimsus * (mW)

Kiirgustihedus * (mW/cm2)

Kogu kestvus (sekundites)

Energia** (Džauli)Standard

970

900

5/10/15/20

4,8J/9,7J/14,5J/19,4J

Tugev võimsus Pluss

1,620

1,500

1/2/3/4

1,6 J/3,2 J/4,8 J/6,5 J

Xtra

2,260

2,100

3

6,8 J

*Jmenovitý výkon ve vzdálenosti 2 mm od hrotu čočky k hornímu povrchu kompozitu.
**Hodnoty v oddíle celková energie na cyklus (jouly) jsou zaokrouhleny na nejbližší desetinu.

Hoiatuste kiirjuhend:
Toitetaseme hoiatus
Vahetage akusid

Temperatuuri hoiatus
Laske seadmel maha jahtuda• Tühi aku: aeglane vilkumine• Väljalülitumine: 3 piiksu, vilkumine• Keelab seadmega töötamise• 3 piiksu• Aeglane vilkumine• Keelab seadmega töötamise

Teenindusega seotud hoiatus

LED-tule hoiatusVõtke seadme remontimiseks ühendust Võtke seadme remontimiseks ühendust
klienditeenindusega
klienditeenindusega• Heli puudub• Vilkumine, 2 sekundit• Võimaldab seadmega töötada

Akude vahetamine/sisestamine:1)
Eemaldage tagakork, keerates seda veerand pööret vastupäeva.
2)
Eemaldage akud.
Sisestage uued patareid plusspool (+) ees.
3)
4)
Kinnitage tagumine kork uuesti, joondades ja vajutades samal ajal päripäeva keerates. Kork klõpsab, kui see on täielikult kinnitunud.
5)
Seade on kasutamiseks valmis.

• 3 pidevat piiksu• Üks kiire vilkumine• Keelab seadmega töötamise

Laadimisaeg ja aku tööiga: Täielikult laetud aku kasutusiga kõvestusvalgustil oleneb režiimist/ajaintervallist, aku tüübist, kasutusmahust ja LED-i tõhususest.
•
Soovitatav laadimisintervall: Laadige vastavalt vajadusele või kui aku tühjenemise indikaator süttib. Oodatav eluiga sõltub kasutamisest.
•
Lisa akud: Kõvendusvalgustiga on kaasas 4 laetavat akut.
Tootjapoolne remont
1)Remonti tohivad teha ainult volitatud teenindused. Ultradent peab andma hoolduspersonalile dokumentatsiooni remonditööde tegemiseks.
2)
Seadmeid remonti, hooldusesse või kalibreerimisse saatmisel eemaldage alati akud kõvestusvalgustist ja laadijast. Tagastuskarpi pakkige akud, laadija, adapter jakõvestusvalgusti eraldi.
3)
Saatke akud vastavalt kohalikele tarneeeskirjadele.
Garantii
Ultradent Products, Inc. ("Ultradent") garanteerib, et see toode peab 5 aasta jooksul alates ostukuupäevast, kui seda kasutatakse tootega kaasasolevate kasutusjuhendite kohaselt, (i)vastama kõigis olulistes aspektides spetsifikatsioonid, mis on toodud tootega kaasas olevas Ultradenti dokumentatsioonis; ja (ii) olema materjali- ja tootmisdefektideta.
Piiratud garantiid ei saa edasi üle kanda ja see kehtib ainult esialgsele ostjale ega laiene toote järgmistele omanikele. See piiratud garantii ei kata muid tarvikuid, nagu, kuid mitte ainult,akud, laadijad, adapterid või adaptiivsed läätsed. See piiratud garantii kaotab kehtivuse, kui toode ebaõnnestub või on kahjustatud hooletuse, kuritarvitamise, väärkasutuse, õnnetuse,muutmise, võltsimise, muutmise või kehtivate kasutusjuhiste mittejärgimise tõttu. See garantii ei hõlma ainult näiteks maha kukkunud ja kahjustatud toodet. Selle piiratud garantii alla
kvalifitseerumiseks tuleb Ultradentile koos defektse tootega esitada ostutõend (nt müügikviitung või muu sarnane dokumentatsioon).
Defektne toode, mis vastab käesolevas dokumendis sätestatud garantiitingimustele, parandatakse või asendatakse Ultradenti äranägemisel. Ultradenti vastutus toote eest ei ületa ühelgi
juhul ostja makstud ostuhinda. Ultradent ei vastuta mitte mingil juhul kaudsete, juhuslike, ettenägematute, ettenägematute, eriliste ega kaudsete kahjude eest, mis tulenevad selle tootekasutamisest või on sellega seotud.

Töötlemine

See toode on ette nähtud kasutamiseks koos kaitseümbrisega. Vaata ülalt hoolduse ja üldpuhastuse kohta.

Ladustamine ja kõrvaldamine

Kui ladustate kõvastamislampi kauem kui 2 nädalat või pakite selle teisaldamiseks kokku, eemaldage alati akud. Kui akud jäetakse pikaks ajaks seadmesse, ilma et neid vahepeal laaditaks,
ei pruugi need enam toimida või neid ei saa enam laadida. Ärge ladustage akusid temperatuuril üle 40 °C (104 °F) või otsese päikesevalguse käes.
Kõvastamislambi ladustamine ja transportimine:• Temperatuur: 10–40 °C (50–104 °F)• Suhteline õhuniiskus: 10–95%• Õhurõhk: 500–1060 hPa
Järgige elektroonikajäätmete (s.t kõvastamislampide, laadijate, akude ja elektritarvikute) kõrvaldamise korral kohalikke jäätmekäitlus- ja töötlemisjuhiseid.

HooldusKõvestusvalgusti üldine puhastamine
Pärast iga kasutamist niisutage marli või pehme lapp heakskiidetud pinnadesinfitseerimisvahendiga ning pühkige pind ja lääts. Volitamata puhastusvahendid võivad kõvendusvalgustit
kahjustada.
SOBIVAD PUHASTUSVAHENDID:•		•		

70% isopropüülalkohol
70% etanool

Valguskilbi üldpuhastus:
VALO Cordless valgusti valguskilpi võib puhastada mis tahes pinnadesinfitseerimisvahendiga. ÄRGE autoklaavige
Parandamine
Kasutaja teostatav hooldus
1)
Kasutage kaitseümbrist, et vältida hambakomposiidi kleepumist läätse pinnale. Vajadusel kasutage kleepunud komposiidi ettevaatlikuks eemaldamiseks plastikust või
154

Tehnilised kaalutlused
LisatarvikudÜksusCE teave

VALO juhtmeta lambi kaitse155ÜksusCE teave

Tehniline teave / andmed

Efektiivsed komposiidi kõvastumise lainepikkuse ribad:

VALO laadija

EFEKTIIVSED KOMPSIITKÕVESTUVAD LAINEPIKKURIBAD(Kasutatav lainepikkuse vahemik: 385-515 nm)

SPEKTRAALKIIRGUSJÕUD (mW/nm)

VALO akud

Efektiivne madalamlainepikkuse riba
395-415 nm

Efektiivne ülemiselainepikkuse riba
440-480 nm

Xtra
Tugev võimsus
Standard

LAINE EST (nm)Atribuut

Teave/andmed

Lääts

Läbimõõt 11,7 mm

Lainepikkusevahemik• Kasutatav lainepikkuse vahemik: 385–515 nm• Tipplainepikkused: 395–415 nm ja 440–480 nm
Nimikiirgavuse võrdlustabelMõõteinst-rument

Valguseintensiivsuse tabel

156‡ Gigaherts-spektrianalüsaator
Kiirgavus

Kogu-võimsusMõõdiku ava

15 mm

15 mm

Standardsevõimsusega režiim
(±10%)

900 mW/cm²

970 mW

High Power Plus(±10%)

1500 mW/cm²

1615 mW

Xtra Power (±10%)2100 mW/cm²

Elektromagnetkiirgus varieerub olenevalt
instrumendi võimalustest, mõõtmismeetodist
ja valguse paigutusest.
‡ Elektromagnetkiirgus vastab standardile ISO
10650, kui seda mõõdetakse Gigahertz-Optik
spektrianalüsaatoriga.

2260 mW

VALO võimas
kõvastamis-lamp

Reitingud: IEC 60601-1 (ohutus), IEC 60601-1-2
(EMÜ)

Kaal:
• Koos akudega: 6 untsi (190 grammi)• Ilma akudeta: 5 untsi (150 grammi)Mõõtmed: (8 × 1,28 × 1,06) tolli, (203 × 32,5 × 27) mm

Laadija toiteallikas

Väljund: 12 V (alalisvool) 500 mA juures
Sisend: 100–240 V (vahelduvvool)
Ultradent P/N 5930 VALO laadija toiteallikas koos
rahvusvaheliste pistikuadapteritega

Reitingud: IEC 60601-1 (ohutus)
Juhtme pikkus: 6 jalga (1,8 meetrit)
VALO laadija toiteallikas on II meditsiiniklassi toiteallikas, mis võimaldab seadme
toitevõrgust isoleerida

157

Mitmesugune teave

Atribuut

Teave/andmed

VALO laadija

VALO 3,6 V (alalisvool) nutikas liitiumraudfosfaataku laadija:• Automaatne väljalülitumine aku täislaadimise korral• Vigaste akude automaatne tuvastamine• Kaitsed: soojuse, ülepinge, lühise, ümberpööratud polaarsuse
⸰ Punane LED-tuli – laadimine
⸰ Roheline LED-tuli – tühi või täis laetud
⸰ LED-tuli ei põle – lühis• Laadimisaeg: 1–3 tundi
Reiting: CE, WEEEVALO akud

Laaditavad: Ohutu liitiumraudfosfaataku (LiFePO4) RCR123A• Tööpinge: 3,2 V (alalisvool)
Reitingud: UL, CE, RoHS, WEEE

Töötingi-mused

Temperatuur: 10–32 °C (50–90 °F)
Suhteline õhuniiskus: 10–95%
Õhurõhk: 700–1060 hPa

Töötsükkel

Kõvastamislamp on mõeldud lühiajaliseks kasutamiseks. Maksimaalse keskkonnatemperatuuri (32 °C) juures järjest kasutamise korral
1 minut SEES, 30 minutit VÄLJAS (jahtumisaeg).

Juhised ja tootja deklaratsioon elektromagnetkiirguse kohtaKõvastamislamp on mõeldud kasutamiseks allpool kirjeldatud elektromagnetilises keskkonnas. Klient või kasutaja peab tagama, et toodet kasutatakse sellises
keskkonnas.
Kiirguskatse

Vastavus

Elektromagnetiline keskkond – juhised

Raadiosageduskiirgus
CISPR 11

1. rühmKõvastamislambis kasutatakse akusid ning seda ei mõjuta elektromagnethäired, raadiosagedus
ega liigpinge piiramine.

Raadiosageduskiirgus
CISPR 11

B-klass

Harmooniline kiirgus
IEC 61000-3-2

Ei kohalduKõvastamislamp kasutab elektri- ja elektromagnetenergiat üksnes sisemiste funktsioonide tarbeks.
Seetõttu on seadmest tuleva raadiosageduskiirguse tase väga madal ja see ei põhjusta tõenäoliselt
lähedalasuvates elektroonikaseadmetes häireid.

Pingekõikumised/väreluskiirgus Ei kohaldu
vastavalt standardile IEC
61000-3-3Harmoonilise kiirguse ja pinge kõikumise katsed ei ole kõvastamislambi puhul kohased, kunaseade töötab akutoitel.
Kõvastamislamp sobib kasutamiseks kõigis hoonetes, k.a elamud ja ruumid, mis on ühendatud
otse avaliku madalpinge vooluvõrguga, millest varustatakse elamuid.

Tõrkeotsing
Kui alltoodud lahendused ei kõrvalda probleemi, võtke ühendust Ultradentiga telefonil 800 552 5512. Väljaspool Ameerika Ühendriike võtke ühendust Ultradenti või
hambaravitoodete edasimüüjaga.
Probleem

Lamp ei lülitu sisseVõimalikud lahendused
1) Vajutage seadme toitesäästurežiimist äratamiseks aja/režiimi või toitenuppu.
2) Vaadake aku laetuse olekut punase aku tühjenemise indikaatori järgi.
3) Veenduge, et seadmesse oleksid õigesti sisestatud uued akud.
4) Kui punane ja kollane hoiatustuli vilguvad, tähendab see, et kõvastamislamp on saavutanud sisemise temperatuuri
ohutuspiiri. Laske kõvastamislambil 10 minutit jahtuda või kasutage seadme kiireks mahajahutamiseks jahedat niisket
käterätti.
5) Kui punane hoiatustuli vilgub ja piiksub pidevalt, võtke seadme remontimiseks ühendust Ultradenti klienditeenindusega.

Lamp ei jää soovitud ajaks sisse.

1) Veenduge režiimi ja ajastamistulede põhjal, et sisestatud aeg oleks õige.
2) Vaadake aku laetuse olekut aku tühjenemise indikaatori järgi.
3) Veenduge, et seadmesse oleksid õigesti sisestatud uued akud.

Lamp ei kõvasta vaike ettenähtud viisil

1) Veenduge, et läätsel poleks kõvastunud vaigu-/komposiidijääke.
2) Veenduge vastavate oranžide UV-kaitseprillide abil, et LED-tuled toimiksid.
3) Kontrollige valgusmõõdikuga toitetaset. Valgusmõõdiku kasutamise korral soovitab Ultradent kontrollidakõvastamislampi standardse võimsusega režiimis.
MÄRKUS. Tegelik arvväärtus on tavaliste valgusmõõdikute ja kõvastamislambi kohandatud LED-komplekti ebatäpsuse tõttu
moonutatud. Valgusmõõdikud on väga erinevad ja need on mõeldud konkreetsete lambi juhtotsakute ja läätsede jaoks.
4) Kontrollige kõvastuva vaigu aegumiskuupäeva.
5) Veenduge, et järgiksite tootja soovitatavat õiget tehnikat.

Akusid ei laeta

1) Veenduge, et akud oleksid õigetpidi laadijasse sisestatud, ja laske akudel 1–3 tundi laadida.
2) Kui laadija punased tuled ei muutu roheliseks, võtke asendusakude ja/või -laadija hankimiseks ühendust Ultradenti
klienditeenindusega.
3) Kui laadijal pole näha ei rohelist ega punast tuld, võtke laadija ja/või vahelduvvooluadapteri tellimiseks või asendamiseksühendust Ultradenti klienditeenindusega.

Laadija ei lae akusid

1) Veenduge, et laadija oleks ühendatud ja vahelduvvooluadapter oleks toimivasse pistikupessa viidud.
2) Kui laadijal pole näha rohelist ega punast tuld, võtke uue laadija ja/või vahelduvvooluadapteri hankimiseks ühendust
Ultradenti klienditeenindusega.

Režiimi või ajaintervalle ei saa muuta

1) Hoidke aja/režiimi nuppu ja toitenuppu korraga all, kuni järjestikused piiksud annavad märku, et kõvastamislamp on
lukust avatud.

Juhised ja tootja deklaratsioon elektromagnetilise häirekindluse kohtaKõvastamislamp on mõeldud kasutamiseks allpool kirjeldatud elektromagnetilises keskkonnas. Klient või kasutaja peab tagama, et toodet kasutatakse sellises
keskkonnas.
Katse tase vastavalt standardile
IEC 60601

Nõuetele vastavuse tase

Elektromagnetilise keskkonna juhised±8 kV kontakt±15 kV õhk±8 kV kontakt±15 kV õhk

Füüsilisele keskkonnale peavad kehtima järgmised
piirangud:1) IP-kood: IP202) Ärge kastke vedelikku.
3) Ärge kasutage kergsüttiva gaasi lähedal. Seadeon mitte-APG ja mitte-AP.
4) Õhuniiskuse vahemik ladustamise korral: 10–95%
5) Temperatuurivahemik ladustamise korral:
10–40 °C

Vastuvõtlikkus kiiretele voolu
muutustele
IEC 61000-4-4±2 kV toiteliinide puhul±1 kV sisend-/väljundliinide puhul±2 kV toiteliinide puhul
1. märkus. Kõvastamislambil
puuduvad I/0 pordid.

Kõvastamislamp töötab akutoitel ja seda ei saa
vahelduvvooluvõrguga ühendada.

Impulssliigpinge
IEC 61000-4-5±1 kV liinist liinini±2 kV liinist maanduseni±1 kV liinist liinini±2 kV liinist maanduseni

Pinge, pingelohud, lühised,toitekat-kestused ja sisendtoiteliinide variatsioonid< 5% U(> 95% U lohk 0,5 tsükli korral)< 5% U(> 95% U lohk 0,5 tsükli korral)

40% U(60% U lohk 5 tsükli korral)

40% U(60% U lohk 5 tsükli korral)

70% U(30% U lohk 25 tsükli korral)

70% U(30% U lohk 25 tsükli korral)< 5% U(> 95% U lohk 5 s korral)< 5% U(> 95% U lohk 5 s korral)
2. märkus. Isetaastumine

30 A/m30 A/mHÄIREKINDLUSE katse
Elektrostaatiline lahendus (ESD)
IEC 61000-4-2

IEC 61000-4-11

Võrgusagedus(50/60 Hz) magnetväli
IEC 61000-4-8

Kuna kõvastamislamp töötab akutoitel, ei rakendu
sellele voolu muutused, impulssliigpinged,
pingelohud, lühised, toitekatkestused ega
vahelduvvooluvõrgu variatsioonid.
Eelnimetatu võib kehtida lisatarvikuna saadaoleva
akulaadija kohta, kuid see on VALO juhtmetakõvastamislambist eraldiseisev seade, mis ei avaldakõvastamislambi talitlusele olulist mõju.
Kui kõvastamislambi akupinge langeb 4 V-ni
(alalisvool), ei saa seadet kasutada. VALO
juhtmeta kõvastamislamp lülitub välja. Uute akude
sisestamise ja ettenähtud toitetaseme taastamise
korral kõvastamislamp taaskäivitub ning naaseb
toitekatkestuse eelsesse olekusse. Kõvastamislamp
taastub ise toitekatkestusest.
Võrgusageduse magnetväljad peaksid vastama
tavapärastele tasemetele, mis esinevad
tavalistes, elu-, koduse tervishoiu, äri-, haigla- ja
militaarkeskkondades.

1. märkus. Kõvastamislambil pole porte või juurdepääsetavaid I/O liine.

158

159

Juhised ja tootja deklaratsioon elektromagnetilise häirekindluse kohta mitte-elushoidmissüsteemide korral

Juhised ja tootja deklaratsioon kaasaskantavate ja mobiilsete raadiosageduslike sideseadmete ning VALO juhtmeta kõvastamislambi vaheliste kauguste kohtaKõvastamislamp on mõeldud kasutamiseks allpool kirjeldatud elektromagnetilises keskkonnas. Klient või kasutaja peab tagama, et toodet kasutatakse sellises
keskkonnas.

Kõvastamislamp on mõeldud kasutamiseks elektromagnetilises keskkonnas, kus raadiosageduskiirguse põhjustatud häired on kontrolli all. Kõvastamislambi
kasutaja saab elektromagnetilisi häireid ennetada, hoides kõvastamislampi kaasaskantavatest ja mobiilsetest raadiosageduslikest sideseadmetest (saatjatest) allpool
soovitatud minimaalsel kaugusel vastavalt sideseadmete maksimaalsele väljundvõimsusele.

HÄIREKIND-LUSE
katse

Katse tase vastavaltNõuetele vastavuse
standardile IEC 60601 tase

Raadio-sagedusejuhtivus

3 Vrms3 VrmsIEC 61000-4-6

150 kHz kuni 80 MHz

150 kHz kuni 80 MHz

Raadiosageduskiirgus

Elektromagnetilise keskkonna juhised
Kaasaskantavaid ja mobiilseid raadiosageduslikke sideseadmeid ei tohi kasutada
kõvastamislambi mis tahes osadele, k.a kaablitele, lähemal kui saatja sageduse kohta
kehtiva võrrandi põhjal arvutatud soovitatav vahemaa.
Soovitatav vahemaa:

80 MHz kuni 800 MHz

800 MHz kuni 2,5 GHz
3 V/m

3 V/m

P on saatja maksimaalne väljundvõimsuse nimiväärtus vattides (W) vastavalt saatja tootja
andmetele ja d on soovitatav vahemaa meetrites (m).
Kohapeal läbiviidud elektromagnetilise uuringu käigus tuvastatud fikseeritudkõrgsageduslike saatjate välja tugevus a peab iga sagedusvahemiku korral olema väiksem
kui ühilduvustase b.

IEC 61000-4-3

80 MHz kuni 2,5 GHz

80 MHz kuni 2,5 GHz

Häired võivad tekkida järgmise sümboliga seadmete lähedal:

Vahemaa vastavalt saatja sagedusele(meetrites)
Saatja maksimaalse väljundvoolu
nimiväärtus
(P vattides)

150 kHz kuni 80 MHz

80 MHz kuni 800 MHz

800 MHz kuni 2,5 GHz

0,01

0,12 meetrit

0,035 meetrit

0,07 meetrit

0,1

0,37 meetrit

0,11 meetrit

0,22 meetrit

1

1,7 meetrit

0,35 meetrit

0,7 meetrit

10

3,7 meetrit

1,11 meetrit

2,22 meetrit

100

11,7 meetrit

3,5 meetrit

7,0 meetritKõvastamislampi on katsetatud standardi IEC 60601-1-2:2014 kohaselt ja sellele on rakendatud kiirgusvälja tugevusega 10 V/m sagedusvahemikus 80 MHz kuni 2,5GHz. Väärtus 3Vrms vastab eespool toodud valemites V1-le ja väärtus 10 V/m E1-le.
Eespool nimetamata maksimaalse väljundvõimsuse nimiväärtusega saatjate puhul saab soovitatavat vahemaad d meetrites (m) hinnata saatja sageduse kohta
kehtivat võrrandit kasutades, kus P on saatja maksimaalse väljundvõimsuse nimiväärtus vattides (W) vastavalt saatja tootja andmetele.
1. MÄRKUS. 80 MHz ja 800 MHz korral rakendub kõrgema sagedusvahemiku vahemaa.
2. MÄRKUS. Need juhised ei pruugi kehtida kõigil juhtudel. Elektromagnetlainete hajumist mõjutavad hoonete, esemete ja inimeste absorptsiooni- ja
peegeldusvõime.

1. MÄRKUS. 80 MHz ja 800 MHz korral rakendub kõrgem sagedusvahemik.
2. MÄRKUS. Need juhised ei pruugi kehtida kõigil juhtudel. Elektromagnetlainete hajumist mõjutavad hoonete, esemete ja inimeste absorptsiooni- ja
peegeldusvõime.
aStatsionaarsete raadiosaatjate, nagu raadiotelefonide (mobiilid/juhtmevabad) põhijaamad ja kaasaskantavad raadiosaatjad, amatöörraadiojaamad, AM-, FM-raadioja telejaamad, väljatugevust pole teoreetiliselt võimalik täpselt prognoosida. Statsionaarsete kõrgsageduslike saatjate elektromagnetilise keskkonna hindamiseksvõiks kaaluda paigalduskoha elektromagnetilist uurimist. Kui mõõdetud väljatugevus asukohas, kus kõvastamislampi kasutatakse, ületab eespool nimetatudkõrgsagedusliku ühilduvuse taset, tuleb VALO juhtmeta kõvastamislampi jälgida, veendumaks, et see toimib ettenähtud viisil. Kui seade töötab ebaharilikult, võib olla
tarvis rakendada lisameetmeid, nagu kõvastamislambi asendi või asukoha muutmine.
b
Sagedusvahemikus 150 kHz kuni 80 MHz peab välja tugevus jääma alla 3 V/m.

160

161

VALO™ GRANDHU

KEZELŐLÁMPA

Termékleírás

Szélessávú spektrumával a VALO Grand polimerizációs lámpát úgy tervezték, hogy az ISO 10650 szerinti 385-515 nm-es hullámhossztartományban minden fényre keményedő terméket
polimerizál. A VALO Grand polimerizációs lámpa az Ultradent VALO újratölthető akkumulátorokkal és akkumulátortöltővel használható. A polimerizációs lámpát úgy tervezték, hogy egy
szabványos fogászati egység konzoljába illeszkedik, és a készlethez mellékelt konzol segítségével egyedileg is felszerelhető.

Nagy VALO termékösszetevők:• 1 – VALO Grand polimerizációs lámpa• 4 – Ultradent VALO újratölthető akkumulátorok• 1 – Ultradent VALO akkumulátortöltő orvosi minőségű 12 VDC/AC hálózati adapterrel• 1 – VALO védőhüvely mintacsomag• 1 – VALO Grand fényvédő pajzs• 1 – Polimerizációs lámpa felületi tartókonzol dupla ragasztószalaggal

Vezérlők áttekintése:

Fénykép

Megjegyzés:• A keresztszennyeződés elkerülése és a kompozit fogászati anyagnak a lencse és a pálcatest felületéhez történő tapadásának megakadályozása érdekében a VALO polimerizációs lámpa
minden egyes használatakor az Ultradent által jóváhagyott védőhüvelyt kell használni. A védőhüvelyek egyszeri használatra készültek

Vezeték nélküli VALO fényvédő:A vezeték nélküli VALO fényvédő ovális alakú, a maximális használhatóság érdekében elforgatható, és átlátszó védőborítóval használható.

Gombok és fények

JelmagyarázatIdőzítési lámpák
Alacsony akkumulátortöltöttség - kijelző
Mód/állapot fényIdő/mód váltógomb

Felhasználás

1) Mindegyik teljesítmény-üzemmód fogászati anyagok fénykezeléssel történő keményítésére használatos. A javasolt kezelési időket lásd a „Gyors üzemmód-útmutatóban”.
MEGJEGYZÉS: A kezelőlámpa úgy van programozva, hogy ciklikusan a standard teljesítményűről a nagy teljesítményű plusz, majd az extra teljesítményű módra lépjen sorban. Ha például
a standard teljesítményű módról az extra teljesítményű módra szeretne váltani, először a nagy teljesítményű plusz módra kell lépni, majd azután az extra teljesítményű módra.
2) A kezelőlámpa tárolja a legutóbb használt időzítési intervallumot és módot, és alapértelmezés szerint visszaállítja ezt a módot, ha megváltozik a mód vagy eltávolítják az elemeket.

Működtetés
KEZELÉSI ÜZEMMÓD: Standard teljesítményű módIDŐZÍTÉSI INTERVALLUMOK: 5, 10, 15, és20 másodperc.
• A kezelőlámpa ebbe az üzemmódba lép alapértelmezés szerint, amikor LEGELŐSZÖR bekapcsolják. A mód/állapot fény zölden világít, és a négy zöld időzítési fény felgyullad astandard teljesítményű módot jelezve.
• Az időzítési intervallum megváltoztatásához nyomja meg gyorsan az idő/mód gombot.
• Nyomja meg a bekapcsológombot a kezelés megkezdéséhez. Ha le szeretné állítani a kezelést az időzítési intervallum vége előtt, nyomja meg újra a bekapcsológombot.

KEZELÉSI ÜZEMMÓD: Nagy teljesítményű plusz mód(Felső)Minden leírt termék használata előtt gondosan olvassa el és értse meg az utasításokat és a biztonsági adatlap információit.

Javallatok/Szándékolt cél

Fotoaktivált fogászati helyreállító anyagok és ragasztók kezelésére szolgáló fényforrás.

Figyelmeztetések és óvintézkedések
2. kockázati csoportVIGYÁZAT: a termék UV-sugárzást bocsát ki. A kitettség a szem vagy a bőr irritációját okozhatja. Alkalmazzon megfelelő védelmet.
VIGYÁZAT: a termék optikai sugárzást bocsát ki, ami veszélyes lehet. Ne nézzen közvetlenül a működő lámpába. Károsíthatja a szemét.
• NE nézzen közvetlenül a polimerizációs lámpába. A betegnek, a klinikai orvosnak és az asszisztenseknek a polimerizációs lámpa használatakor minden esetben borostyán színű
szemvédőt kell használniuk.
• Az áramütés veszélyének elkerülése érdekében a berendezésen nem végezhetők módosítások. Csak a mellékelt Ultradent VALO tápegységet és csatlakozóadaptereket használja. Ha
valamelyik alkatrész megsérült, ne használja azt, hanem hívja az Ultradent ügyfélszolgálatát, és rendelje meg a cserealkatrészt.
• A 30 cm-nél (12 hüvelyknél) közelebb használt hordozható rádiófrekvenciás kommunikációs berendezések ronthatják az eszköz teljesítményét.
• A helytelen működés, az elektromágneses sugárzás növekedésének és az elektromágneses zavartűrés csökkenésének elkerülése érdekében csak engedélyezett tartozékokat,kábeleket és tápegységeket használjon (olvassa el az Elektromágneses sugárzás című fejezetet).
• Az akkumulátorok kezelésével összefüggő elektromos tűz keletkezésének elkerülése érdekében:⸰ Az újratölthető akkumulátorokat NE használja nem újratölthető akkumulátorokkal vagy más típusú akkumulátorokkal.
⸰ A nem újratölthető akkumulátorokat NE próbálja meg feltölteni. A nem újratölthető akkumulátorok töltése sérülést vagy anyagi kárt okozhat.
⸰ NE autoklávozza és ne permetezze be folyadékkal az akkumulátorokat, az akkumulátor érintkezőit, a töltőt és a hálózati adaptert. Ha az akkumulátortöltő érintkezőin a korrózió
jeleit észleli, hívja az Ultradent ügyfélszolgálatát, és rendelje meg a cseretartozékot.
⸰ Az akkumulátorokat NE töltse gyúlékony anyagok közelében.
⸰ A töltőt NE a klinikai helyiségben tárolja.
• A sérülések elkerülése érdekében NE használjon korrodált (rozsdás), horpadt, szagot vagy folyadékot kibocsátó, szakadt vagy hiányzó csomagolású vagy más módon sérült
akkumulátorokat. Csereakkumulátor rendeléséhez hívja az Ultradent ügyfélszolgálatát.
• Csak az ajánlott akkumulátorokat használja. Az alternatív akkumulátorok hibás működést eredményezhetnek.
• A termikus irritáció vagy sérülések elkerülése érdekében kerülje a közvetlenül egymás után következő kikeményítési ciklusokat, és semmilyen üzemmódban ne tegye ki a száj
lágyrészeit 10 másodpercnél tovább a fénynek. Ha hosszabb a megszilárdulási idő, a lágyszövetek felmelegedésének elkerülése érdekében használjon több kikeményítési ciklust, aciklusok között iktasson be pihenőidőt, vagy használjon kettős polimerizációs terméket.
• Körültekintően kezelje azokat a betegeket, akik nemkívánatos fotobiológiai reakciókban vagy érzékenységben szenvednek, akik kemoterápiás kezelés alatt állnak, vagy akik
fényérzékenyítő gyógyszereket szednek.
• Ez az egység érzékeny lehet az erős mágneses vagy statikus elektromos mezőkre, amelyek megzavarhatják a programozást. Ha úgy gondolja, hogy ilyen esemény történt, az egységet
húzza ki egy pillanatra a konnektorból, majd dugja vissza a konnektorba.
• A polimerizációs lámpát NE tisztítsa maró vagy súroló hatású tisztítószerekkel, ne tegye autoklávba, és ne merítse bele semmilyen ultrahangos fürdőbe, fertőtlenítőszerbe,
162

Lépésenkénti utasításokElőkészítés1) A polimerizációs lámpa használata előtt fel az akkumulátorokat (olvassa el az Akkumulátor karbantartása című részt).
2) Minden használat előtt helyezzen egy új védőhüvelyt a polimerizációs lámpára, amely biztosítja a lehető legjobb eredmény elérését, és minimalizálja a gyűrődéseket a lencsén.

Bekapcsológomb(ok)(Felső vagy alsó)(Alsó)

tisztítóoldatba vagy folyadékba. A leírt utasítások be nem tartása esetén a polimerizációs lámpa működésképtelenné válhat.
• A berendezés károsodásának elkerülése érdekében NE dugja be az ujjait, valamint ne helyezzen különböző eszközöket és egyéb tárgyakat a polimerizációs lámpa
akkumulátorrekeszébe.
• A berendezés károsodásának elkerülése érdekében NE próbálja meg megtisztítani az arany érintkezőket vagy az akkumulátorrekesz bármely részét. Ha aggálya van, hívja az Ultradentügyfélszolgálatát.
• A keresztszennyeződés kockázatának megelőzése érdekében a védőhüvelyek egyetlen betegen használhatók.
• A korrózió veszélyének csökkentése érdekében használat után távolítsa el a védőhüvelyt.
• A nem megfelelő kikeményedés elkerülése érdekében ne használja a polimerizációs lámpát, ha a lencse megsérült.

IDŐZÍTÉSI INTERVALLUMOK: 1, 2, 3, és 4 másodperc.
• Standard teljesítményű módban tartsa 2 másodpercig lenyomva az idő/mód váltógombot, majd engedje el. A mód/állapot fény sárgán világít, és a négy zöld időzítési fény felgyulladés villog a nagy teljesítményű módot jelezve.
• Az időzítési intervallum megváltoztatásához nyomja meg gyorsan az idő/mód gombot.
• Nyomja meg valamelyik bekapcsológombot a kezelés megkezdéséhez. Ha le szeretné állítani a kezelést az időzítési intervallum vége előtt, nyomja meg újra valamelyik
bekapcsológombot.
• A standard teljesítményű módba való visszatéréshez tartsa 2 másodpercig lenyomva az idő/mód váltógombot, majd engedje el, ez átviszi az extra teljesítményű módba. Tartsa újra 2
másodpercig lenyomva, majd engedje el. A mód/állapot fény zölden világít, és a négy zöld időzítési fény felgyullad a standard módot jelezve.

KEZELÉSI ÜZEMMÓD: Extra teljesítményű módIDŐZÍTÉSI INTERVALLUM: Csak 3 másodperc (Megjegyzés: Az extra teljesítményű módnak van egy 2 másodperces biztonsági késleltetése minden kezelési ciklus végén, hogy korlátozza
a felmelegedést egymást követő kezelések során. A késleltetés végén sípolás jelzi, hogy a készülék készen áll a további használatra.)• Standard teljesítményű módból tartsa 2 másodpercig lenyomva az idő/mód váltógombot, engedje el, tartsa újra 2 másodpercig lenyomva, és engedje el. A mód/állapot fény sárgán
villog, és három zöld időzítési fény felgyullad és villog az extra teljesítményű módot jelezve.
• Nyomja meg a bekapcsológombot a kezelés megkezdéséhez. Ha le szeretné állítani a kezelést az időzítési intervallum vége előtt, nyomja meg újra a bekapcsológombot.
• A standard teljesítményű módba való visszatéréshez tartsa 2 másodpercig lenyomva az idő/mód váltógombot, majd engedje el. A mód/állapot fény zölden világít, és a zöld időzítési
fények felgyulladnak a standard teljesítményű módot jelezve.

Alvó üzemmód: A kezelőlámpa 60 másodperc inaktivitás után ALVÓ üzemmódba kerül, amit a mód/állapot fény lassú villogása jelez. A készülék felemelése vagy megérintése felébreszti akezelőlámpát, és automatikusan visszatér az utolsó beállításra. Az elem élettartamának maximalizálása érdekében hagyja békén a kezelőlámpát, ha nem használja..

Tisztítás

1) Minden beteg után dobja a használt védőborítót a szemétbe.
2) Lásd a „Kezelés” című fejezetet.

A tartóval kapcsolatos utasítások

1) A tartót sima, olajmentes felületre kell szerelni.
2) Tisztítsa meg a felületet alkoholos bedörzsöléssel.
3) Húzza le a tartó ragasztószalagját.
4) Helyezze úgy a tartót, hogy a kezelőlámpát felfelé emelve lehessen kivenni. Nyomja erősen a helyére.

163

Gyors üzemmód-útmutató:
Standard teljesítmény
1000 mW/cm2

Üzemmód

Nagy teljesítmény plusz
1600 mW/cm2

Extra teljesítmény
3200 mW/cm2

Bekapcsológomb
Mód/időzítés LED-ek

Időzítési gombokIdőzítési opciók

5s

10s

15s

20s

1s

2s

3s

4s

csak 3sIdőzítés változtatásához

Nyomja meg és engedje el gyorsan az időzítési gombot, hogy pörgesse az időzítési opciókat.

Mód változtatásához

Tartsa 2 másodpercig lenyomva az időzítési gombot, majd engedje el. A kezelőlámpa a következő módra ugrik.

Jelmagyarázat

Világító LED-ek

Teljesítmény * (mW)

Villogó LED-ek

Besugárzás * (mW/cm2)

Teljes expozíciós idő (másodperc)

Energia** (Joule)Standard

970

900

5/10/15/20

4,8 J/9,7 J/14,5 J/19,4 J

High Power Plus

1620

1500

1/2/3/4

1,6 J/3,2 J/4,8 J/6,5 J

Xtra

2260

2100

3

6,8 J

*Névleges kimeneti teljesítmény a lencse hegyétől a kompozit felső felületéig mért 2 mm-es távolságban.
**A ciklusonkénti összes energia (joule) szakaszban szereplő értékek a legközelebbi tizedesjegyre vannak kerekítve.

Gyors útmutató a figyelmeztetésekről:Teljesítményszint figyelmeztetésHőmérséklet figyelmeztetés

Szerviz figyelmeztetés

LED figyelmeztetésCserélje ki az elemeket

Hagyja lehűlni

Javításért forduljon azügyfélszolgálathoz

Javításért forduljon azügyfélszolgálathoz• Alacsony akkumulátortöltés:lassan villog• Kikapcsolás: 3 sípolás, villogás• Letiltja a működést

• 3 sípolás• Lassú villogás• Letiltja a működést

• Nincs hang
• 2 másodpercenként villog• Engedi a működést

Az akkumulátorok töltése:1) A töltőt és az akkumulátorokat olyan helyen tárolja, ahol nem érintkezhetnek folyadékkal vagy fertőtlenítő spray-kkel.
2) Az akkumulátorokat úgy kell behelyezni a töltőbe, hogy a pozitív (+) vég a töltőn található jelzőfények felé mutasson.
3) A zöld fények azt jelzik, hogy az akkumulátorok készen állnak a használatra.
4) Az akkumulátorok feltöltése 1-3 órát vesz igénybe. Az akkumulátorokat csak akkor vegye ki a töltőből, ha azok teljesen feltöltött állapotba kerültek.
MEGJEGYZÉS: Ha az akkumulátorok három óránál hosszabb ideje vannak a töltőben, és a töltőn található piros lámpa nem vált zöldre, az akkumulátor sérült, és nem tölthető. Használjon
másik akkumulátort, vagy hívja az Ultradent ügyfélszolgálatát, és rendeljen egy újat.
Az akkumulátorok cseréje/behelyezése:
1) Fordítsa el a hátó kupakot egy negyed fordulattal az óramutató járásával ellentétes irányba, és vegye le azt.
2) Távolítsa el az akkumulátorokat.
3) Az új akkumulátorokat a pozitív (+) oldalukkal előre helyezze be.
4) Helyezze vissza a hátsó kupakot, óvatosan nyomja meg, és forgassa el az óramutató járásával megegyező irányba. Ha a kupak a helyén van, kattanó hangot fog hallani.
5) Az egység készen áll a használatra.

Quick Curing Guide:
Kikeményítési mód

Javítás
Felhasználó által végzett karbantartás
1) Annak megakadályozására, hogy a fogászati kompozit hozzátapadjon a lencse felületéhez, használjon védőhüvelyt. Ha a letapadt kompozit eltávolítására van szükség, használjon
műanyag vagy rozsdamentes acél fogászati műszert, és a műveletet óvatosan végezze el. Ne használjon olyan eszközöket, amelyek károsítják a lencsét.
2) A fénymérők nagymértékben különböznek egymástól, és meghatározott fényvezető csúcsokhoz és lencsékhez készültek. Az Ultradent azt javasolja, hogy rendszeresen ellenőrizze
a kimenetet Standard Power módban. MEGJEGYZÉS: a hagyományos fénymérők és a polimerizációs lámpa egyedi LED-egységének pontatlansága miatt a valódi numerikus kimenet
elferdülAz akkumulátorok karbantartása
Az akkumulátorok töltése és cseréje
A polimerizációs lámpa 4 újratölthető lítium-vasfoszfát akkumulátorral van felszerelve. Az Ultradent kizárólag a VALO újratölthető akkumulátorok használatát javasolja.

• Folyamatos 3 sípolás• Egyetlen gyors villanás• Letiltja a működést

Alacsony az akkumulátor töltöttségi szintje: A polimerizációs lámpa az alacsony töltöttséget jelző lámpa piros villogásával jelzi a felhasználónak, hogy az akkumulátor cseréjére vanszükség. Ha az akkumulátor töltöttségi szintje túl alacsony lesz, a lámpa 3 sípoló hangjelzéssel figyelmezteti a felhasználót, hogy az egység csak az akkumulátorom feltöltése vagy az új
akkumulátorok behelyezése után lesz ismét használható. (Olvassa el a figyelmeztetéseket tartalmazó rövid útmutatót)
Töltési idő és az akkumulátor élettartama: A polimerizációs lámpa teljesen feltöltött akkumulátorának élettartama az Üzemmód/Idő beállítástól, az akkumulátor típusától, a használatmértékétől és a LED hatékonyságától függ.
• Javasolt töltési intervallum: Az akkumulátort szükség esetén vagy az alacsony töltöttségi szintet jelző fény kigyulladásakor kell feltölteni. A várható élettartam a használattól függ.
• Extra akkumulátorok: A polimerizációs lámpához 4 újratölthető akkumulátor tartozik.
GaranciaAz Ultradent Products, Inc. („Ultradent”) szavatolja, hogy ez a termék a vásárlás időpontjától számított 5 évig a termékhez mellékelt használati utasításnak megfelelően üzemeltetve (i)minden lényeges szempontból megfelel az Ultradent által a termékhez mellékelt dokumentációban meghatározott specifikációknak; és (ii) anyag- és gyártási hibáktól mentes lesz.
Ez a korlátozott garancia nem átruházható, és kizárólag az eredeti vásárlóra vonatkozik, a termék későbbi tulajdonosaira nem terjed ki. Ez a korlátozott garancia nem terjed ki semmilyen
más tartozékra, például az akkumulátorokra, a töltőkre, az adapterekre és az adaptív lencsékre nem terjed ki. Ez a korlátozott garancia érvényét veszti, ha a termék gondatlanság,
visszaélés, helytelen használat, baleset, módosítás, manipuláció, változtatás vagy a vonatkozó használati utasítások be nem tartása miatt hibásodik meg vagy sérül meg. Példaként
említjük, hogy a leejtésből adódó sérülésekre nem vonatkozik a garancia. A korlátozott garancia érvényesítéséhez az Ultradentnek a hibás termékkel együtt a vásárlást igazoló bizonylatot(pl. nyugta vagy hasonló dokumentum) is el kell küldeni.
Az Ultradent az itt meghatározott garanciális feltételeknek megfelelő hibás termék esetén saját belátása szerint dönt a termék javításáról vagy cseréjéről. Az Ultradent felelőssége atermékért semmilyen esetben sem haladhatja meg a vásárló által kifizetett vételárat. Az Ultradent semmilyen körülmények között nem vállal felelősséget a termék használatából eredő
vagy azzal kapcsolatban álló közvetett, véletlen, előre látható, előre nem látható, egyedi vagy utólagos károkért.

Kezelés
Ezt a terméket védőhüvellyel történő használatra tervezték. A karbantartásról és az általános tisztításról fent olvashat. Tárolás és ártalmatlanítás
Ha a kezelőlámpát 2 hétnél hosszabb ideig tárolja vagy utazáshoz becsomagolja, mindig vegye ki az elemeket. Ha az elemeket hosszabb ideig hagyja a készülékben feltöltés nélkül,előfordulhat, hogy működésképtelenné válnak vagy nem tölthetők újra. Ne tárolja az elemeket 40 °C (104 °F) -t meghaladó hőmérsékleten vagy közvetlen napfényben.
Kezelőlámpa tárolása és szállítása:• Hőmérséklet: +10 °C - +40 °C (+50 °F - +104 °F)• Relatív páratartalom: 10% - 95%• Környezeti nyomás: 500 - 1060 hPa
Elektronikus hulladékok (azaz kezelőlámpák, töltők, elemek és tápegységek) ártalmatlanításakor kövesse a helyi hulladék- és újrahasznosítási irányelveket.

Karbantartás
Javítás

A polimerizációs lámpa általános tisztítása
Minden használat után nedvesítsen meg egy gézt vagy puha ruhát jóváhagyott felületfertőtlenítővel, majd törölje le a felületet és a lencsét. A nem jóváhagyott tisztítószerek károsíthatják
a polimerizációs lámpát.
JÓVÁHAGYOTT TISZTÍTÓSZEREK:• 70%-os izopropil-alkohol• 70%-os etanol

A fényvédő pajzs általános tisztítása:
Tisztítsa meg a VALO Vezeték nélküli fényvédő pajzsot valamilyen felületfertőtlenítővel. NE autoklávozzon.
164

Technikai megfontolások
Tartozékok
Cikk

CE adatok

Vezeték nélküli VALO fényvédő

165

Cikk

CE adatok

Technikai információk/adatok

Tényleges kompozitkezelési hullámhosszsávok:VALO töltőHATÉKONY KOMPOZIT KEMÉNYEZŐ HULLÁMHOSSZSÁVOK
(Használható hullámhossz-tartomány: 385-515 nm)
SPEKTRÁLIS SUGÁRZÁS ERŐ (mW/nm)

VALO elemek

Hatékony alsó
hullámhossz sáv
395-415 nm

Hatékony felső
hullámhossz-sáv
440-480 nm

Xtra
Nagy teljesítmény
StandardHULLÁM ENGTH (nm)Jellemző

Információ/specifikáció

LencseÁtmérő 11,7 mm

Hullámhossz¬tartomány• Használható hullámhossztartomány: 385-515 nm• Csúcshullámhosszak: 395-415 nm és 440-480 nm
Névleges sugárzáskibocsátási összehasonlító táblázatMérőműszer

Fény-intenzitások
táblázata166‡ Gigahertz-es spektrumanalizátor
Kibo-csátásÖssztelje-sítmény

A mérő apertúrája

15 mm

15 mm

Standard
teljesítmény(±10%)

900 mW/cm²

970 mW

Nagy teljesítmény
plusz (±10%)1500 mW/cm²

1615 mW

Extra teljesítmény(±10%)

2100 mW/cm²

2260 mW

A kibocsátott sugárzás a műszer
képességétől, a mérési módszertől és a fényelhelyezkedésétől függően változik.
‡ Gigahertz-Optik spektrumanalizátorral mérvea kibocsátott sugárzás megfelel az ISO 10650
szabványnak.

Nagy VALO
kezelőlámpa

Osztályozások: IEC 60601-1 (biztonság), IEC 60601-1-2(EMC)

Tömeg:• Elemekkel: 6 oz. (190 gramm)• Akkumulátor nélkül: 5 oz. (150 gramm)
Méret: 203 x 32,5 x 27 mm

Töltő tápegység

Kimenet - 12 V DC 500 mA-nél
Bemenet - 100 - 240 V AC
Ultradent cikkszám: 5930, VALO töltő tápegység
nemzetközi dugókkal

Osztályozások: IEC 60601-1 (biztonság)
Vezeték hossza - 1,8 méter
A VALO töltő tápegység orvosi II. osztályú tápegység, és szigetelést biztosít a hálózati
tápellátástól

167

Jellemző

Információ/specifikáció

VALO töltő

VALO 3,6 V DC lítium-vasfoszfát intelligens akkumulátortöltő:• Automatikus kikapcsolás, ha teljesen fel van töltve
• Hibás akkumulátorok automatikus felismerése• Védelmek: hő, túlterhelés, rövidzárlat, fordított polaritás
⸰ Piros LED – Töltés
⸰ Zöld LED – Üres vagy teljesen feltöltve
⸰ LED kikapcsolva – rövidzárlat
• Töltési idő: 1-3 óra
Osztályozás: CE, WEEE

Vegyes információk
Irányelvek és gyártói nyilatkozat - Elektromágneses kibocsátás
A kezelőlámpát a lent meghatározott elektromágneses környezetben történő felhasználásra szánták. A vevőnek vagy a felhasználónak kell biztosítania az ilyen
környezetben történő használatot.

Újratölthető: biztonságos kémiai lítium-vas-foszfát (LiFePO4) RCR123A• Üzemi feszültség: 3,2 VDC, 400 mAH (szükséges wattóra)
Besorolások: UL 62133-2, CE, RoHS, WEEEA cellák a kapacitás 70%-ának elérése előtt >2000 töltési/kisütési ciklusra vannak tervezve. Az akkumulátorokat 3-5 évente célszerű kicserélni, deaz élettartam a használat függvényében változhat. Szükség esetén cserélje ki.

VALO elemekHőmérséklet: +10 - +32 °CMűködési feltételek Relatív páratartalom: 10% - 95%
Környezeti nyomás: 700 - 1060 hPaMűködési ciklus

Megfelelőség

Elektromágneses környezet - irányelvek

Rádiófrekvenciás kibocsátás
CISPR 11

1. csoport

A kezelőlámpa elemeket használ, és nem érinti az EMI, RF vagy túlfeszültség-elnyomás.

Rádiófrekvenciás kibocsátás
CISPR 11

B osztályú

A kezelőlámpa csak a belső funkcióihoz használ elektromos és elektromágneses energiát. Ezért
a rádiófrekvenciás kibocsátás nagyon alacsony, és valószínűleg nem okoz interferenciát a közeli
elektronikus berendezésekben.

Harmonikus kibocsátás IEC
61000-3-2

n.a.

Harmonikus kibocsátási és feszültségingadozás-tesztelés nem alkalmazható a kezelőlámpára, mert
elemmel működik.

Feszültségingadozások/villogás
IEC 61000-3-3

n.a.

A kezelőlámpa minden létesítményben való használatra alkalmas, ideértve a hazai létesítményeket
és azokat, amelyek közvetlenül a háztartási használatra szánt épületeket ellátó kisfeszültségű
hálózathoz csatlakoznak.

A kezelőlámpa rövid időtartamú működtetésre szolgál. A maximális környezeti hőmérsékleten (32 °C) 1 perc folyamatos bekapcsolás után 30
percre ki kell kapcsolni (lehűlési időtartam).
Irányelvek és gyártói nyilatkozat - Elektromágneses immunitás

Hibakeresés
Ha az alább javasolt megoldások nem oldják meg a problémát, kérjük, hívja az Ultradent-et a 800-552-5512-es telefonszámon. Az Egyesült Államokon kívül hívja az
Ultradent forgalmazót vagy a fogászati kereskedőt.
Probléma

168

Kibocsátási teszt

Lehetséges megoldások

A lámpa nem gyullad fel

1) Az energiatakarékos módból való kilépéshez nyomja meg az idő/mód vagy a bekapcsológombot..
2) Ellenőrizze az elemek töltöttségi állapotát a piros alacsony akkumulátortöltés jelzőfénnyel.
3) Ellenőrizze, hogy a friss elemeket megfelelően helyezte-e be a készülékbe.
4) Ha a piros és sárga figyelmeztető LED-ek villognak, az azt jelenti, hogy a kezelőlámpa elérte a belső hőmérséklet
biztonsági határértékét. Hagyja lehűlni a kezelőlámpát 10 percig vagy egy hűvös, nedves törülközővel gyorsan hűtse lea készüléket.
5) Ha a piros figyelmeztető LED villog és folyamatosan sípol, hívja fel javításért az Ultradent ügyfélszolgálatát.

A lámpa nem marad felgyújtva akívánt ideig1) Ellenőrizze a mód és időzítési fényeket, hogy helyes időt adott-e meg.
2) Ellenőrizze az elemek töltöttségi állapotát az alacsony akkumulátortöltés jelzőfénnyel.
3) Ellenőrizze, hogy a friss elemeket megfelelően helyezte-e be a készülékbe.

A lámpa nem kezeli megfelelően agyantákat

1) Ellenőrizze a lencsét, nem maradt-e rajta megkeményedett gyanta/kompozit.
2) Megfelelő borostyánsárga UV-védőszemüveg használatával ellenőrizze, hogy a LED-fények működnek-e.
3) Ellenőrizze a teljesítményszintet fénymérővel. Ha fénymérőt használ, az Ultradent azt javasolja, hogy a kezelőlámpát
standard teljesítményű módban ellenőrizze.
MEGJEGYZÉS: A valódi numerikus kijelzést torzíthatja a közönséges fénymérők pontatlansága és a kezelőlámpa által használt
egyedi LED-csomag. A fénymérők nagyban különböznek egymástól, és speciális fényvezető csúcsokhoz és lencsékhez vannak
tervezve.
4) Ellenőrizze a gyógygyanta szavatossági idejét.
5) Győződjön meg róla, hogy a gyártó ajánlásának megfelelő technikát követ.

Az elemek nem töltődnek

1) Győződjön meg róla, hogy az elemek a megfelelő helyzetben vannak-e behelyezve a töltőbe, és hagyja az elemeket 1-3órán keresztül töltődni.
2) Ha a töltő piros fényei nem változnak zöldre, hívja az Ultradent ügyfélszolgálatát, hogy csereelemeket és/vagy töltőt
rendeljen.
3) Ha a töltőn sem a zöld, sem a piros fény nem látható, hívja az Ultradent ügyfélszolgálatát a töltő és/vagy a váltóáramú
adapter megrendelése vagy cseréje érdekében.

A töltő nem tölti az elemeket1) Győződjön meg róla, hogy a töltő csatlakoztatva van, és a hálózati adapter csatlakoztatva van egy működő hálózati
aljzathoz.
2) Ha a töltőn a zöld vagy a piros fény nem látható, hívja az Ultradent ügyfélszolgálatát új töltőért és/vagy váltóáramú
adapterért

Nem lehet változtatni a módot vagy azidőintervallumot

1) Tartsa lenyomva mind az idő/mód, mind a bekapcsológombot, amíg egy sor sípolás jelzi, hogy a kezelőlámpa zárolása
fel van oldva.

A kezelőlámpát a lent meghatározott elektromágneses környezetben történő felhasználásra szánták. A vevőnek vagy a felhasználónak kell biztosítania az ilyen
környezetben történő használatot.
IMMUNITÁSI teszt

IEC 60601 tesztelési szint

Megfelelőségi szint

Elektromágneses környezet - irányelvek

Elektrosztatikus kisülés (ESD)± 8 kV érintkezéssel
± 15 kV levegőn keresztül± 8 kV érintkezéssel
± 15 kV levegőn keresztül

A fizikai környezetet a következőkre kell korlátozni:
1) IP kód: IP20
2) Ne mártsa folyadékba.
3) Ne használja gyúlékony gáz közelében. Az egység
nem APG és nem AP.
4) Tárolási páratartalom-tartomány: 10% - 95%
5) Tárolási hőmérséklet-tartomány: 10 - 40 °C

Gyors villamos tranziens/burst
IEC 61000-4-4± 2 kV a tápegység vezetékeire± 1 kV a bemeneti/kimeneti
vezetékekre± 2 kV a tápegység vezetékeire
1. megjegyzés: A kezelőlámpán
nincs I/O port

A kezelőlámpa elemmel működik, és nem képes
csatlakozni a váltóáramú tápfeszültséghez.

Túlfeszültség
IEC 61000-4-5± 1 kV vezetékek között
± 2 kV vezeték és föld között

± 1 kV vezetékek között
± 2 kV vezeték és föld között

Feszültség, esések, rövidzárlatok,megszakítások és változások atápegység bemeneti vezetékein

< 5% U(> 95% U esés 0,5 cikluson át)< 5% U(> 95% U esés 0,5 cikluson át)

40% U(60% U esés 5 cikluson át)

40% U(60% U esés 5 cikluson át)

70% U(30% U esés 25 cikluson át)

70% U(30% U esés 25 cikluson át)< 5% U(> 95% U esés 5 másodpercen át)< 5% U(> 95% U esés 5 másodpercen át)
2. megjegyzés: önállóan helyreáll

30 A/m30 A/mIEC 61000-4-2

IEC 61000-4-11

Tápellátási frekvencia(50/60 Hz) mágneses mezője
IEC 61000-4-8

Mivel a kezelőlámpa elemmel működik, nemérintik az elektromos tranziensek, túlfeszültségek,feszültségcsökkenések, rövidzárlatok, megszakításokvagy a váltóáramú tápfeszültség változásai.
A tartozék akkumulátortöltőre vonatkozhatnak afentiek, de külön van, és nem kritikus a vezeték nélküli
VALO kezelőlámpa működéséhez.
Ha a kezelőlámpa elemfeszültsége 4 V DC-recsökken, a készülék nem működtethető. A vezeték
nélküli VALO kezelőlámpa kikapcsol. Amikor új
elemeket helyeznek be, és a megfelelő tápfeszültség
helyreáll, a kezelőlámpa újraindul, és visszatér afeszültségkimaradás előtti állapotba. A kezelőlámpaönállóan helyreáll feszültségkimaradás esetén.
A tápellátási frekvencia mágneses mezőjének a tipikus
lakó-, otthoni, egészségügyi, kereskedelmi, kórházi
vagy katonai környezet tipikus helyére jellemző szintenkell lennie.

1. megjegyzés: A kezelőlámpa nem rendelkezik porttal vagy hozzáférhető I/O vonalakkal.

169

Irányelvek és gyártói nyilatkozat - Elektromágneses immunitás - Nem életfenntartó rendszerek
A kezelőlámpát a lent meghatározott elektromágneses környezetben történő felhasználásra szánták. A vevőnek vagy a felhasználónak kell biztosítania az ilyen
környezetben történő használatot.
IMMUNITÁSI teszt

IEC 60601 tesztelési
szint

Megfelelőségi szint

Vezetési
rádiófrekvencia

3 Vrms3 VrmsIEC 61000-4-6

150 kHz - 80 MHz

150 kHz - 80 MHz

Elektromágneses környezet - irányelvek
frekvenciájára vonatkozó egyenlet alapján kiszámított javasolt elkülönítési távolság.
Javasolt elkülönítési távolság:

80 - 800 MHz

800 MHz - 2,5 GHz
Sugárzott
rádiófrekvencia

3 V/m

3 V/m

P a sugárzó eszköz maximális leadott teljesítménye Watt (W) mértékegységben a gyártó
adatai alapján, és d a javasolt elkülönítési távolság méterben.
Az elektromágneses helyszíni felmérésa által meghatározott rögzített rádiófrekvenciás
sugárzó eszközök térerejének kisebbnek kell lennie, mint az egyes frekvenciatartományokb
megfelelőségi szintje.

IEC 61000-4-3

80 MHz - 2,5 GHz

80 MHz - 2,5 GHzInterferencia következhet be az alábbi jelekkel ellátott eszközök esetében:1. MEGJEGYZÉS: a 80 MHz és 800 MHz frekvencia esetén a nagyobb frekvenciatartomány érvényes.
2. MEGJEGYZÉS: ezek az irányelvek nem érvényesek minden helyzetre. Az elektromágneses hullámok terjedésére hatással van az épületek, tárgyak és emberek általi
elnyelődés és visszaverődés.
a
A rögzített sugárzó eszközök - például (mobil/vezeték nélküli) rádiótelefonok és földi mobil rádiók bázisállomásai, szárazföldi rádiók, amatőr rádiók, AM és
FM rádiók, valamint televíziós műsorszórók - térerejét elméletben nem lehet pontosan megjósolni. A rögzített rádiófrekvenciás sugárzó eszközök által okozott
elektromágneses környezet felméréséhez meg kell fontolni elektromágneses helyszíni felmérést. Ha a mért térerő azon a helyen, ahol a kezelőlámpát használják,
meghaladja a fenti rádiófrekvenciás megfelelőségi szintet, a vezeték nélküli VALO kezelőlámpát meg kell figyelni, hogy ellenőrizzék a normál működést. Ha
rendellenes működést tapasztalnak, további intézkedésekre lehet szükség, például a kezelőlámpa átirányítására vagy áthelyezésére.
b
A 150 kHz - 80 MHz frekvenciatartományon túl a térerőnek kisebbnek kell lennie, mint 3 V/m.

170

Irányelvek és gyártói nyilatkozat - Javasolt elkülönítési távolságok a hordozható és mobil rádiófrekvenciás kommunikációs berendezések és a vezeték nélküli VALO
kezelőlámpa között
A kezelőlámpát javasolt olyan elektromágneses környezetben használni, ahol a rádiófrekvenciás sugárzási zavarokat kontrollálják. A kezelőlámpa használója segíthet
megelőzni az elektromágneses interferenciát, ha a hordozható és mobil rádiófrekvenciás kommunikációs berendezések és a kezelőlámpa között betartja a lent
feltüntetett távolságokat, figyelembe véve a kommunikációs berendezés maximális leadott teljesítményét.

Névleges maximális leadottteljesítmény
sugárzó eszköz(P, Watt)

Elkülönítési távolság a sugárzó eszköz frekvenciája alapján(méter)150 kHz – 80 MHz

80 MHz – 800 MHz

800 MHz – 2,5 GHz

0,01

0,12 méter

0,035 méter

0,07 méter

0,1

0,37 méter

0,11 méter

0,22 méter

1

1,7 méter

0,35 méter

0,7 méter

10

3,7 méter

1,11 méter

2,22 méter

100

11,7 méter

3,5 méter

7,0 méter

A kezelőlámpát az IEC 60601-1-2:2014-nek megfelelően tesztelték, és a sugárzott térerő 10 V/m-nél kisebb értékkel megfelelt 80 Mhz és 2,5 GHz között. A fenti
képletekben a 3 Vrms érték felel meg a V1-nek és a 10 V/m érték felel meg az E1-nek.
Azoknál a sugárzóknál, amelyekhez nem adták meg a lent látható maximális leadott teljesítményt, a javasolt d elkülönítési távolság méterben (m) a sugárzó
frekvenciáját alkalmazó képlet alapján határozható meg, ahol a P a sugárzó eszköz maximális leadott teljesítménye Watt (W) mértékegységben a gyártó adatai
alapján.
1. MEGJEGYZÉS: a 80 MHz és 800 MHz frekvencia esetén a nagyobb frekvenciatartomány elkülönítési távolsága érvényes.
2. MEGJEGYZÉS: ezek az irányelvek nem érvényesek minden helyzetre. Az elektromágneses hullámok terjedésére hatással van az épületek, tárgyak és emberek általi
elnyelődés és visszaverődés.

171

VALO™ GRAND

LVCIETĒŠANAS GAISMAS
Produkta apraksts

Ar platjoslas spektru VALO Grand cietināšanas lampa ir paredzēta visu ar lampu cietināmo produktu polimerizācijai viļņu garuma diapazonā no 385 līdz 515 nm atbilstoši ISO 10650. VALOGrand cietināšanas lampa izmanto Ultradent VALO uzlādējamās baterijas un bateriju lādētāju. Cietināšanas lampa ir paredzēta standarta zobārstniecības ierīces kronšteinā, vai arī to varuzstādīt pēc pasūtījuma, izmantojot komplektā iekļauto kronšteinu.

VALO lielā produktu komponenti:
•
•
•
•
•
•

Fotoattēls

1) Pirms cietināšanas lampas izmantošanas uzlādējiet baterijas (skatiet sadaļu "Bateriju apkope)."2) Pirms katras lietošanas reizes uzvelciet jaunu barjeras uzmavu virs cietināšanas lampas, līdz minimumam samazinot lēcas krokas, lai iegūtu labākos rezultātus.

VALO bezvadu gaismas aizsargs:

VALO bezvadu gaismas aizsargs ir ovālas formas, to var pagriezt, lai izmantotu maksimāli, un to var izmantot ar caurspīdīgu aizsargmehānismu.

Izmantot
Pogas un gaismasLeģendaLaika gaismas
Zems akumulators - indikatorsRežīma / statusa indikators
Laiks / režīma maiņas poga(Tops)

Visiem aprakstītajiem produktiem pirms lietošanas rūpīgi izlasiet un saprotiet visas instrukcijas un SDS informāciju.

Lietošanas indikācijas / nolūks

Apgaismojuma avots fotoaktivizētiem zobu atjaunojošiem materiāliem un līmvielām.

1) Katru jaudas režīmu izmanto zobārstniecības materiālu apstrādei ar foto ierosinātājiem. Ieteicamās sacietēšanas reizes skatiet Ātrā režīma rokasgrāmatā.
PIEZĪME: Cietēšanas gaisma ir ieprogrammēta, lai secīgi pārietu no standarta strāvas uz Augstas-jaudas Plus un uz Ekstra jaudas režīmu. Piemēram, lai pārslēgtos no standarta strāvasrežīma uz Ekstra jaudas režīmu, ir jāpārvietojas augstas-jaudas Plus režīmā un pēc tam uz Ekstra jaudas režīmu.
2) Cietēšanas gaisma saglabā pēdējo izmantoto laika intervālu un režīmu, un tas pēc noklusējuma atgriezīsies, kad tiek mainīti režīmi vai ja baterijas ir izņemtas.

Darbība
Sacietēšanas Režīms: Standarta jaudas režīms

LAIKA INTERVALS: 5, 10, 15, 20 unsekundes.
• Sildīšanas gaisma noklusē šo režīmu, kad tā ir sākotnēji ieslēgta. Režīms / stāvokļa gaisma būs zaļā krāsā un izgaismosies četri zaļie laika indikatori, norādot standarta strāvas režīmu.
• Lai mainītu laika intervālus, ātri nospiediet laika / režīma pogu.
• Lai ārstētu, nospiediet pogu Jauda. Lai pārtrauktu sacietēšanu pirms laika intervāla pabeigšanas, vēlreiz nospiediet pogu Jauda.

SACIETĒŠANAS REŽĪMS : Augstas jaudas plus režīms

LAIKA INTERVALS: 1, 2, 3, un4 sekundes.
• Standarta strāvas režīmā nospiediet un turiet laika / režīma maiņas pogu 2 sekundes un atlaidiet. Režīms / statusa indikators būs oranžs, un četri zaļie laika indikatori izgaismosies un
mirgo, norādot Augstas Jaudas režīmu.
• Lai mainītu laika intervālus, ātri nospiediet laika / režīma pogu.
• Lai ārstētu, nospiediet pogu Jauda. Lai pārtrauktu sacietēšanu pirms laika intervāla pabeigšanas, vēlreiz nospiediet jebkuru strāvas padeves pogu.
• Lai atgrieztos pie standarta strāvas režīma, nospiediet un turiet laika / režīma maiņas pogu 2 sekundes un atlaidiet, un tas būs cikls uz Ekstra Jaudas režīmu. Nospiediet un turiet vēl 2sekundes un atlaidiet. Režīms / stāvokļa indikators būs zaļš un četri zaļie laika indikatori izgaismosies, norādot standarta režīmu.

SACIETĒŠANAS REŽĪMS: Ekstra jaudas režīms

Brīdinājumi un piesardzības pasākumi
Riska grupa 2
UZMANĪBU UV starojums no šī produkta. Iedarbības rezultātā var rasties acu vai ādas kairinājums. Izmantojiet atbilstošu aizsargu.
UZMANĪBU Iespējams, ka šī produkta radītais bīstamais optiskais starojums. Neskatieties uz darba lampas. Var kaitēt acīm.
• NESKATIETIES tieši uz gaismas izvadi. Pacientam, ārstam un asistentam vienmēr jālieto dzeltenas krāsas UV acu aizsarglīdzekļi, kad tiek izmantota cietināšanas lampa.
• Lai novērstu elektriskās strāvas trieciena risku, šī aprīkojuma modifikācijas nav atļautas. Izmantojiet tikai komplektācijā iekļauto Ultradent VALO barošanas avotu un spraudņa
adapterus. Ja šie komponenti ir bojāti, nelietojiet tos un sazinieties ar Ultradent klientu apkalpošanas dienestu, lai pasūtītu nomaiņu.
• Pārnēsājamo RF sakaru iekārtu veiktspēja var pasliktināties, ja to izmanto tuvāk par 30 cm (12 collām).
• Izmantojiet tikai autorizētus piederumus, kabeļus un barošanas avotus, lai novērstu nepareizu darbību, palielinātu elektromagnētisko emisiju vai samazinātu elektromagnētisko
imunitāti (skatiet sadaļu "Elektromagnētiskā emisija").
• Lai izvairītos no elektriskā aizdegšanās riska, kas saistīts ar bateriju apstrādi:⸰ NEjaunojiet uzlādējamās baterijas ar neuzlādējamām baterijām vai cita veida baterijām.
⸰ NEmēģiniet uzlādēt neuzlādējamās baterijas. Neuzlādējamo bateriju uzlāde var izraisīt savainojumus vai īpašuma bojājumus.
⸰ NEautoklavējiet un nesmidziniet baterijas, bateriju kontaktus, lādētāju vai maiņstrāvas barošanas adapteri ar jebkāda veida šķidrumu. Ja uz bateriju lādētāja kontaktiem parādās
korozija, zvaniet Ultradent klientu apkalpošanas dienestam, lai pasūtītu nomaiņu.
⸰ NEuzlādējiet baterijas viegli uzliesmojošu materiālu tuvumā.
⸰ NEglabājiet lādētāju klīniskajā operāciju telpā.
• Lai izvairītos no savainošanās riska, NElietojiet baterijas, kas ir ar koroziju (rūsu), iespiedušās, izdala smaku vai šķidrumus, kurām ir saplēsts iesaiņojums vai tā nav, vai tie ir citādi
bojāti. Zvaniet uz Ultradent klientu apkalpošanas dienestu, lai pasūtītu rezerves baterijas.
• Izmantojiet tikai ieteiktās baterijas. Alternatīvas baterijas var izraisīt darbības traucējumus.
• Lai novērstu termiska kairinājuma vai ievainojumu risku, izvairieties no cietināšanas cikliem un nepakļaujiet mutes mīkstos audus tiešā tuvumā ilgāk par 10 sekundēm jebkurā režīmā.
Ja nepieciešams ilgāks cietināšanas laiks, izmantojiet vairākus cietināšanas ciklus ar atpūtas periodiem starp cikliem vai izmantojiet divkāršās cietināšanas līdzekli, lai izvairītos nomīksto audu sildīšanas.
• Esiet piesardzīgi, ārstējot pacientus, kuri cieš no nevēlamām fotobioloģiskām reakcijām vai jutīguma, pacientus, kuriem tiek veikta ķīmijterapija, vai pacientus, kuri tiek ārstēti arfotosensibilizējošiem medikamentiem.
• Šī iekārta var būt jutīga pret spēcīgiem magnētiskiem vai statiskiem elektriskiem laukiem, kas var traucēt programmēšanu. Ja jums ir aizdomas, ka tas ir noticis, uz brīdi atvienojiet
ierīci un pēc tam atkal pievienojiet to kontaktligzdai.
• NEnoslaukiet cietināšanas lampu ar kodīgiem vai abrazīviem tīrīšanas līdzekļiem, neautoklavējiet vai neiegremdējiet jebkāda veida ultraskaņas vannā, dezinfekcijas līdzekļos, tīrīšanas
šķīdumā vai šķidrumā. Iekļauto apstrādes instrukciju neievērošana var padarīt cietināšanas lampu nedarbojamu.
172

Pakāpeniski norādījumi
Sagatavošana• Lai novērstu savstarpēju piesārņojumu un novērstu zobu kompozītmateriāla pielipšanu lēcas un zižļa korpusa virsmai, katru reizi VALO cietināšanas lampai ir jāizmanto Ultradentapstiprināta barjeras uzmava. Barjeras uzmavas ir paredzētas lietošanai vienam pacientam.

Barošanas poga (-as)(Augšā vai apakšā)(Apakšā)Lai nesabojātu iekārtu, NEievietojiet pirkstus, instrumentus vai citus priekšmetus cietināšanas lampas bateriju nodalījumā.
Lai nesabojātu iekārtu, NEmēģiniet tīrīt zelta kontaktus vai jebkuru bateriju nodalījuma daļu. Ja rodas bažas, zvaniet Ultradent klientu apkalpošanas dienestam.
Lai novērstu savstarpējas inficēšanās risku, barjeras uzmavas ir paredzētas vienam pacientam.
Lai samazinātu korozijas risku, pēc lietošanas noņemiet barjeras uzmavu.
Lai samazinātu nepietiekami sacietējušu sveķu risku, neizmantojiet cietināšanas lampu, ja lēca ir bojāta.

Piezīme:1 – VALO Grand cietināšanas lampa
4 – Ultradent VALO uzlādējamās baterijas
1 – Ultradent VALO bateriju lādētājs ar medicīnisku 12 V līdzstrāvas maiņstrāvas adapteri
1 – VALO Barrier Uzmavu paraugu iepakojums
1 – VALO Grand gaismas vairogs
1 – Cietināšanas lampas montāžas kronšteins ar dubultu līmlenti

Kontroles pārskats:

•
•
•
•
•

LAIKA INTERVĀLS: Tikai 3 sekundes (Piezīme: Ekstra Jaudas režīmā katras sacietēšanas cikla beigās ir 2 sekunžu drošības aizkave, lai ierobežotu sildīšanu secīgas sacietēšanas laikā.
Aizkavēšanās beigās skaņas signāls norāda, ka ierīce ir gatava lietošanai tālāk).
• Standarta strāvas režīmā nospiediet laika / režīma maiņas pogu 2 sekundes, atlaidiet, nospiediet un turiet vēl 2 sekundes un atlaidiet. Režīms / statusa indikators būs oranžs un
mirgos, un trīs no zaļajiem laika indikatoriem iedegsies un mirgos, norādot Ekstra Jaudas režīmu.
• Lai ārstētu, nospiediet pogu Jauda. Lai pārtrauktu sacietēšanu pirms laika intervāla pabeigšanas, vēlreiz nospiediet ieslēgšanas pogu.
• Lai atgrieztos standarta strāvas režīmā, nospiediet un 2 sekundes turiet nospiestu laika / režīma pogu un atlaidiet. Režīms / statusa indikators būs zaļš un iedegsies zaļš laika
indikators, norādot parastā režīma režīmu.

Miega režīms:Cietēšanas gaisma ieslēgsies miega režīmā pēc 60 sekundēm bezdarbības, kā to norāda režīma / statusa gaismas lēnā mirgošana. Ierīces paņemšana vai pieskaršanās
modinās cietēšanas gaismu un automātiski darbosies pēdējie izmantotie iestatījumi. Lai palielinātu akumulatora kalpošanas laiku, atstājiet cietēšanas gaismu netraucēti, kad tonelietojat.

Satīrīt

1) Pēc katra pacienta izmetiet izmantotās barjeras uzmavas pie standarta atkritumiem.
2) Skatiet sadaļu Pārstrādes sadaļa.

Montāžas kronšteinu instrukcijas

1) Kronšteins jāuzstāda uz līdzenas, ne-eļļainas virsmas.
2) Notīriet virsmu ar berzes alkoholu.
3) Nogrieziet pamatnes līmlenti.
4) Novietojiet kronšteinu tā, lai noņemšanas brīdī sacietēšanas gaisma paceltos uz augšu. Novietojiet stingri vietā.

173Ātrā režīma ceļvedis:
Standarta jauda
1000 mW/cm2

RežīmsAugstas jaudas plus
1600 mW/cm2

Ekstra Jauda
3200 mW/cm2

Ieslēgšanas / izslēgšanas pogaRežīms / laika indikatori

Kā uzlādēt baterijas:
1) Novietojiet lādētāju un baterijas vietā, kur tie nesaskaras ar šķidrumiem vai dezinfekcijas līdzekli.
2) Ievietojiet baterijas lādētājā ar pozitīvo (+) galu pret lādētāja indikatora gaismām.
3) Zaļās gaismas norāda, ka baterijas ir gatavas lietošanai.
4) Bateriju uzlāde prasīs 1-3 stundas. Atstājiet baterijas lādētājā, līdz tās ir gatavas lietošanai.
PIEZĪME. Ja lādētāja sarkanā gaisma neiedegas zaļā krāsā, kad baterijas ir lādētas ilgāk par trim stundām, baterija var būt bojāta un to nevar uzlādēt. Izmēģiniet jaunu bateriju vai
zvaniet Ultradent klientu apkalpošanas dienestam, lai pasūtītu jaunu uzlādējamo bateriju komplektu.

Laika pogas

Laika iespējas

5s

10s

15s

20s

1s

2s

3s

4s

Tikai 3s

Lai mainītu laiku

Nospiediet un atlaidiet laika pogu, lai ātri pārvietotos pa laika iespējām.

Lai mainītu režīmus

Nospiediet un turiet laika taustiņu 2 sekundes un atlaidiet. Ārstēšanas gaismas ciklā uz nākamo režīmu.

LeģendaCietās gaismas diodes

Mirgojošas gaismas diodes

Cietināšanas režīms

Jauda * (mW)

Izstarojums * (mW/cm2)

Kopējais ekspozīcijas laiks (sekundēs)Enerģija** (Džouli)

Standart

970

900

5/10/15/20

4.8J/9.7J/14.5J/19.4J

High Power Plus

1,620

1,500

1/2/3/4

1.6J/3.2J/4.8J/6.5J

Xtra

2,260

2,100

3

6.8J

*Nominālā jauda 2 mm attālumā no objektīva gala līdz kompozīta augšējai virsmai.
**Vērtības kopējās enerģijas ciklā (džoulos) sadaļā ir noapaļotas līdz tuvākajai desmitdaļai.

Jaudas līmeņa brīdinājums
Nomainiet baterijas• Zems akumulatora uzlādes līmenis:
lēni mirgo• Izslēgt: 3 pīkstieni, mirgo• Aizliedz darbību

Temperatūras brīdinājums

Servisa brīdinājums

LED brīdinājums

Ļaujiet atdzist

Zvaniet uz klientu apkalpošanas
dienestu remontam

Zvaniet uz klientu apkalpošanas
dienestu remontam• 3 pīkstieni• Lēni mirgo• Aizliedz darbību• Nav skaņas
• Mirgo ik pēc, 2 sekundēm• Ļauj darboties

Kā nomainīt/ievietot baterijas:
1) Noņemiet aizmugurējo vāciņu, pagriežot pretēji pulksteņrādītāja virzienam par ceturtdaļu apgrieziena.
2) Izņemiet baterijas.
3) Ievietojiet jaunas baterijas ar pozitīvo (+) pusi vispirms.
4) Uzlieciet atpakaļ aizmugurējo vāciņu, izlīdzinot un viegli spiežot, vienlaikus griežot pulksteņrādītāja virzienā. Vāciņš noklikšķinās, kad tas ir pilnībā piestiprināts.
5) Ierīce ir gatava lietošanai.
Zems bateriju līmenis: Cietināšanas lampa norāda lietotājam, ka ir laiks nomainīt baterijas, kad Zema baterijas uzlādes līmeņa indikators mirgo sarkanā krāsā. Ja baterijas uzlādes līmeniskļūst pārāk zems, atskanēs 3 pīkstienu brīdinājuma skaņa, un cietināšanas lampa neļaus turpmāku darbību, kamēr baterijas nav uzlādētas vai nav ievietotas jaunas baterijas. (Skatiet ātrobrīdinājumu rokasgrāmatu)Ātrās sacietēšanas rokasgrāmata:Ātrās brīdināšanas rokasgrāmata:1) Izmantojiet barjeras uzmavu, lai novērstu zobu kompozītmateriālu pielipšanu lēcas virsmai. Ja nepieciešams, izmantojiet plastmasas vai nerūsējošā tērauda zobārstniecības
instrumentu, lai uzmanīgi noņemtu visus pielipušos kompozītmateriālus. Neizmantojiet instrumentus, kas sabojās objektīvu.
2) Gaismas mērītāji ir ļoti atšķirīgi un ir paredzēti īpašiem gaismas vadu uzgaļiem un lēcām. Ultradent iesaka regulāri pārbaudīt izvadi Standart Power režīmā. PIEZĪME: patiesā skaitļu
izvade būs sagrozīta parasto gaismas mērītāju neprecizitātes un pielāgotā LED komplekta cietināšanas lampā dēļ.
Bateriju apkope
Bateriju uzlāde un maiņa
Cietināšanas lampai ir 4 atkārtoti uzlādējamas litija dzelzs fosfāta baterijas. Ultradent iesaka izmantot tikai VALO uzlādējamās baterijas.

• Nepārtraukti 3 pīkstieni• Viena ātra zibspuldze• Aizliedz operācijas

Uzlādes laiks un baterijas darbības laiks: Pilnībā uzlādētas baterijas darbības laiks cietināšanas lampā ir atkarīgs no režīma/laika intervāla, baterijas veida, lietošanas daudzuma un LED
efektivitātes.
•Ieteicamais uzlādes intervāls: Uzlādējiet pēc vajadzības vai, kad iedegas Zema baterijas uzlādes līmeņa indikators. Darbības ilgums ir atkarīgs no lietošanas veida.
•
Papildu baterijas: Cietināšanas lampai ir 4 atkārtoti uzlādējamas baterijas.
Ražotāja remonts
1) Remontu drīkst veikt tikai pilnvarots servisa personāls. Ultradent nodrošina apkalpojošo personālu ar dokumentāciju remontdarbu veikšanai.
2) Nosūtot ierīces remontam, apkopei vai kalibrēšanai, vienmēr izņemiet baterijas no cietināšanas lampas un lādētāja. Atgriešanas kastē atsevišķi ietiniet baterijas, lādētāju, adapteri uncietināšanas lampu.
3) Nosūtiet baterijas saskaņā ar vietējiem noteikumiem.
Garantija
Ultradent Products, Inc. ("Ultradent") garantē, ka šis produkts 5 gadus no iegādes datuma, ja tas tiek izmantots saskaņā ar produkta komplektācijā iekļautajām lietošanas instrukcijām, (i)visos būtiskajos aspektos atbilst specifikācijām, kas noteiktas produktam pievienotajā Ultradent dokumentācijā; un (ii) jābūt bez materiāla un ražošanas defektiem.
Šī ierobežotā garantija nav nododama tālāk un attiecas tikai uz sākotnējo pircēju un neattiecas uz turpmākajiem produkta īpašniekiem. Šī ierobežotā garantija neattiecas uz citiem
piederumu komponentiem, piemēram, bet ne tikai, akumulatoriem, lādētājiem, adapteriem vai adaptīvajām lēcām. Šī ierobežotā garantija tiek anulēta, ja izstrādājums nedarbojas vai
ir bojāts nolaidības, ļaunprātīgas izmantošanas, nepareizas lietošanas, negadījuma, modifikācijas, iejaukšanās, pārveidošanas vai piemērojamo lietošanas instrukciju neievērošanas
dēļ. Piemēram, šī garantija neattiecas uz izstrādājumu, kas ir nomests un bojāts. Lai pretendētu uz šo ierobežoto garantiju, uzņēmumam Ultradent kopā ar bojāto produktu ir jāiesniedz
pirkuma apliecinājums (piemēram, pārdošanas čeks vai līdzīga dokumentācija).
Bojāts produkts, kas atbilst šeit izklāstītajiem garantijas nosacījumiem, pēc Ultradent ieskatiem tiks salabots vai nomainīts. Ultradent atbildība par produktu nekādā gadījumā
nepārsniedz pirkuma cenu, ko samaksājis pircējs. Ultradent nekādā gadījumā nav atbildīgs par jebkādiem netiešiem, nejaušiem, neparedzētiem, neparedzētiem, īpašiem vai izrietošiem
bojājumiem, kas radušies no šī produkta lietošanas vai saistībā ar to.

Apstrāde

Šis produkts ir paredzēts lietošanai ar barjeras uzmavu. Apkopi un vispārējo tīrīšanu skatiet iepriekš.

Uzglabāšana un atbrīvošanās

Ja uzglabājat sacietēšanas gaismu ilgāk par 2 nedēļām vai iesaiņojiet brauciena laikā, vienmēr izņemiet baterijas. Ja baterijas paliek ierīcē ilgu laiku bez uzlādēšanas, tās var kļūtnefunkcionālas vai neuzlādējamas. Neuzglabājiet baterijas temperatūrā virs 40°C (104°F) C vai tiešos saules staros.

Uzturēšana

Cietināšanas lampas vispārējā tīrīšana
Pēc katras lietošanas reizes samitriniet marli vai mīkstu drāniņu ar apstiprinātu dezinfekcijas līdzekli un noslaukiet virsmu un lēcu. Neatļauti tīrīšanas līdzekļi var sabojāt cietināšanas
lampu.
PIEŅEMAMIE TĪRĪŠANAS LĪDZEKĻI:• 70% izopropilspirts• 70% etanols

Gaismas vairoga vispārējā tīrīšana:Notīriet VALO Cordless gaismas vairogu, izmantojot jebkuru virsmas dezinfekcijas līdzekli. NEautoklāvējiet.
Remonts
Lietotāja veikta apkope
174Ārstēšanas gaismas Uzglabāšana un transportēšana:• Temperatūra: no + 10 ° C līdz + 40 ° C (+ 50 ° F līdz + 104 ° F)
• Relatīvais mitrums: 10% līdz 95%• Apkārtējais spiediens: 500 hPa līdz 1060 hPa
Atbrīvojoties no elektroniskajiem atkritumiem (t.i., ārstēšanas gaismas, lādētāji, baterijas un barošanas avoti), ievērojiet vietējās atkritumu un pārstrādes vadlīnijas.

Tehniskie apsvērumi
Piederumi

175Vienums

CE informācija

Tehniskā informācija / dati

Efektīvas salikšanas viļņa garuma joslas:

VALO Bezvadu Gaismas Aizsargs

SPEKTRĀLĀ STAROJUMA JAUDA (mW/nm)

EFEKTĪVAS KOMPOZĪTA CIETĒJOŠAS VIĻŅU GARUMA JOSLAS(Izmantojamais viļņa garuma diapazons: 385-515 nm)

VALO lādētājs

VALO baterijas

Efektīva zemākaviļņa garuma josla
395-415 nm

Efektīvā augšējā viļņa
garuma josla
440-480 nm

XtraHigh PowerStandard

VILNIS ENGTH (nm)Atribūts

Informācija / specifikācija

Objektīvs

Diametrs 11,7 mm

Viļņu garuma
diapazons• Izmantojamā viļņa garuma diapazons: 385 - 515 nm• Maksimālais viļņa garums: 395 - 415 nm un 440 - 480 nm
Nominālā starojuma izejas salīdzinājuma diagramma
Mērīšanas
instruments

Gaismasintensitātes tabula

VALO Lielā
Cietēšanas Gaisma‡ Gigaherta spektra analizators
Izeja

Kopējā jauda

Meter diafragmas
atvērums

15 mm

15 mm

Standarta jauda(±10%)

900 mW/cm²

970 mW

Augstas jaudas plus(±10%)

1500 mW/cm²

1615 mW

Ekstra jauda (±10%)2100 mW/cm²

Novērtējumi: IEC 60601-1 (Drošība), IEC 60601-1-2(EMC)

Izeja - 12VDC pie 500mA
Lādētāja barošanas Ieeja - 100VAC līdz 240VAC
avots
Ultradent P / N 5930 VALO lādētāja barošanas avots arstarptautiskiem spraudņu ieliktņiem

176

Starojuma izplūde mainīsies atkarībā noinstrumenta iespējām, mērīšanas metodes un
gaismas izvietojuma.
‡ Starojuma izplūde atbilst ISO 10650, mērotar Gigahertz-Optik spektra analizatoru.

2260 mW
Svars:• Ar baterijām: 6 oz. (190 grami)• Bez baterijām: 5 oz. (150 grami)
Izmēri: (8 x 1,28 x 1,06) collas (203 x 32,5 x 27) mm
Novērtējumi: IEC 60601-1 (Drošība)Vada garums - 1,8 m
VALO lādētāja barošanas avots ir II klases medicīniskās klases barošanas avots un
nodrošina izolāciju no MAINS jaudas

177Dažāda informācija

Atribūts

Informācija / specifikācija

Vadlīnijas un Ražošanas deklarācija par elektromagnētiskajām emisijām

VALO lādētājs

VALO 3.6VDC litija dzelzs fosfāta viedais lādētājs:• Automātiska izslēgšanās, kad tas ir pilnībā uzlādēts
• Bojātu bateriju automātiska noteikšana• Aizsardzība: Termiskā, pārslodze, īssavienojums, pretēja polaritāte⸰ Sarkans LED - uzlāde⸰ Zaļā LED - tukša vai pilnībā uzlādēta
⸰ LED izslēgts - īssavienojums• Uzlādes laiks: 1 - 3 stundas
RVērtējums: CE, EEIA

Emisiju tests

Atbilstība

Elektromagnētiskā vide – norādījumi

RF emisijas
CISPR 11

1. grupa

Cietēšanas gaisma izmanto baterijas, un to neietekmē EMI, RF vai pārsprieguma novēršana.

RF emisija
CISPR 11

B klaseCietēšanas gaisma izmanto elektrisko un elektromagnētisko enerģiju tikai to iekšējām funkcijām.
Tāpēc jebkuras RF emisijas ir ļoti zemas un, visticamāk, netraucēs tuvumā esošās elektroniskās
iekārtas.

VALO baterijas

Uzlādējams: Droša ķīmija litija dzelzs fosfāts (LiFePO4) RCR123A
Darba spriegums: 3.2VDC, 400mAH (nepieciešams vatstundas)•
Vērtējumi: UL 62133-2, CE, RoHS, WEEE
Elementi ir paredzēti >2000 uzlādes/izlādes cikliem, pirms tie sasniedz 70% no to jaudas. Baterijas ieteicams nomainīt ik pēc 3-5 gadiem,taču kalpošanas laiks var atšķirties atkarībā no lietošanas. Nomainiet pēc vajadzības.

Harmoniskās emisijas
IEC 61000-3-2

N/A

Harmoniskās emisijas un sprieguma svārstību testēšana nav piemērojama cietēšanas gaismai, jotā ir uzlādēta ar akumulatoru.

Darbībasnosacījumi

Temperatūra: no + 10 ° C līdz + 32 ° C (+ 50 ° F līdz + 90 ° F)
Relatīvais mitrums: 10% līdz 95%
Apkārtējais spiediens: 700 hPa līdz 1060 hPa

Sprieguma svārstības /mirgošana IEC
61000-3-3

N/ACietēšans gaisma ir piemērota lietošanai visās iestādēs, tostarp mājsaimniecībās un tajās, kas
ir tieši pieslēgtas publiskajam zemsprieguma elektrotīklam, kas apgādā ēkas mājsaimniecības
vajadzībām.

Cikls

Cietēšanas gaisma ir paredzēta īslaicīgai darbībai. Maksimālā apkārtējās vides temperatūrā (32 ° C) 1 minūte ieslēgtam ciklam, 30 minūtes
izslēgtam ciklam (atdzesēšanas periods).
Vadlīnijas un Ražošanas deklarācija par elektromagnētisko imunitāti

Problēmu novēršana
Ja turpmāk piedāvātie risinājumi problēmu nenovērš, lūdzu, zvaniet uz Ultradent pa tālruni 800.552.5512. Ārpus Amerikas Savienotajām Valstīm izsauciet savu Ultradentizplatītāju vai zobārstniecības pārstāvi.
Problēma

178

Cietēšanas gaisma ir paredzēta izmantošanai turpmāk norādītajā elektromagnētiskajā vidē. Klientam vai lietotājam būtu jānodrošina, ka tas tiek izmantots šādā vidē.

Iespējamie risinājumi

Gaisma neieslēdzas

1) Nospiediet laika / režīma vai strāvas padeves taustiņu, lai pamodinātu ierīci no enerģijas taupīšanas režīma.
2) Pārbaudiet, vai akumulatora uzlādes stāvoklī ir sarkans zems akumulatora uzlādes līmenis.
3) Pārbaudiet, vai ierīcē ir pareizi ievietotas jaunas baterijas.
4) Ja sarkanās un dzeltenās brīdinājuma gaismas diodes mirgo, tas nozīmē, ka cietēšanas gaisma ir sasniegusi iekšējo
temperatūras drošības robežu. Ļaujiet sacietēšanas gaismai atdzist 10 minūtes vai izmantojiet aukstu, mitru dvieli, laiātri atdzesētu ierīci.
5) Ja sarkanais brīdinājuma LED mirgo un nepārtraukti pīkst, izsauciet Ultradent klientu apkalpošanas dienestu.

Gaisma nepaliek uz vēlamo laiku

1) Pārbaudiet režīmu un laika indikatorus, lai pareizi ievadītu laiku.
2) Pārbaudiet akumulatora uzlādes stāvokļa indikatoru.
3) Pārbaudiet, vai ierīcē ir ievietotas jaunas baterijas.

Gaisma neārstē sveķus pareizi

1) Pārbaudiet objektīvu, lai iegūtu atlikušos cietinātos sveķus / kompozītus.
2) Izmantojot atbilstošu dzintarkrāsas UV acu aizsardzības ierīci, pārbaudiet, vai darbojas LED gaismas.
3) Pārbaudiet jaudas līmeni ar gaismas mērītāju. Ja lietojat gaismas mērītāju, Ultradent iesaka pārbaudīt sacietēšanas
gaismu standarta jaudas režīmā.
PIEZĪME: Patiesā skaitliskā izeja būs novirzīta parasto gaismas mērierīču neprecizitātes dēļ un pielāgotā gaismas diodes
iepakojuma dēļ. Gaismas mērītāji ir ļoti atšķirīgi un tie ir paredzēti īpašiem gaismas vadiem.
4) Pārbaudiet derīguma termiņu uz cietēšanas sveķiem.
5) Pārliecinieties, vai tiek ievērota pareiza tehnika saskaņā ar ražotāja ieteikumiem.

Baterijas netiek uzlādētas

1) Pārliecinieties, vai akumulatori ir ievietoti lādētājā pareizi, un ļaujiet akumulatoriem uzlādēties 1-3 stundas.
2) Ja lādētājā sarkanā gaisma nemainās uz zaļo, zvaniet uz Ultradent klientu apkalpošanas dienestu, lai pasūtītu rezervesbaterijas un / vai lādētāju.
3) Ja uz lādētāja nav redzama ne zaļa, ne sarkana gaisma, sazinieties ar Ultradent klientu apkalpošanas dienestu, lai pasūtītu
vai nomainītu lādētāju un / vai maiņstrāvas adapteri.

Lādētājs neuzlādē baterijas

1) Pārliecinieties, ka lādētājs ir pievienots un maiņstrāvas adapteris ir pievienots darba strāvas kontaktligzdai.
2) Ja uz lādētāja nav redzama zaļa vai sarkana gaisma, sazinieties ar Ultradent klientu apkalpošanas dienestu, lai saņemtu
jaunu lādētāju un / vai maiņstrāvas adapteri.

Nevar mainīt režīmu vai laika intervālus1) Turiet gan laika, gan režīma un barošanas pogas uz leju, līdz sērijveida pīkstieni norāda, ka cietēšanas gaisma ir
atbloķēta.

Cietēšanas gaisma ir paredzēta izmantošanai turpmāk norādītajā elektromagnētiskajā vidē. Klientam vai lietotājam būtu jānodrošina, ka tas tiek izmantots šādā vidē.
IMMUNITĀTES tests

IEC 60601 testa līmenis

Atbilstības līmenis

Elektromagnētiskās vides vadība

Elektrostatiskā izlāde (ESD)± 8 kV kontakts± 15 kV gaisa± 8 kV kontakts± 15 kV gaisa

Fiziskā vide jāierobežo šādi:1) IP kods: IP20
2) Nemērciet šķidrumā.
3) Nelietot ap degošām gāzēm. Vienība ir bez APG
un bez AP.
4) Uzglabāšanas mitruma diapazons: 10% - 95%
5) Uzglabāšanas temperatūras diapazons: 10 °C - 40 ° C

Elektrisks ātrs pārejošs / pārrauts
IEC 61000-4-4± 2 kV barošanas līnijām± 1 kV ieejas / izejas līnijām± 2 kV barošanas līnijām
1. piezīme: Cietēšanas gaismas
nav I / 0 porti

Cietēšanas gaisma ir uzlādēta ar akumulatoru unnevar pieslēgties maiņstrāvas AVOTAM.

SPārspriegums
IEC 61000-4-5± 1 kV līnija līdz līnijai± 2 kV līnija uz zemi

± 1 kV līnija līdz līnijai± 2 kV līnija uz zemi

Spriegums, kritumi, īsumi,
elektroenerģijas padeves līniju
pārtraukumi un izmaiņas.

<5% U(> 95% kritums U 0,5 ciklam)<5% U(> 95% kritums U 0,5 ciklam)

40% U(60% iemērkšana U 5 ciklos)

40% U(60% iemērkšana U 5 ciklos)

70% U(30% iemērkšana U uz 25 cikliem)

70% U(30% iemērkšana U uz 25 cikliem)<5% U(> 95% iemērkšana U uz 5
sekundēm)<5% U(> 95% iemērkšana U uz 5
sekundēm)
2. piezīme: Pašatgūšanās

30 A/m30 A/mIEC 61000-4-2

IEC 61000-4-11

Jaudas frekvence(50/60 Hz) magnētiskais lauks
IEC 61000-4-8

Tā kā cietēšanas gaisma ir uzlādēta ar akumulatoru, tānav pakļauta elektriskām pārejām, pārspriegumiem,sprieguma kritumiem, īsumiem, pārtraukumiem vaimaiņstrāvas STRĀVAS jaudas izmaiņām.
Papildu akumulatora lādētājs var būt pakļauts
iepriekšminētajam, taču tas ir nošķirts un nav kritisks
VALO Bezvadu cietēšanas gaismas darbībai.
Ja cietēšanas gaismas akumulatora spriegums nokrītas
līdz 4VDC, ierīce neļauj darboties. VALO Bezvadu
cietēšanas gaisma izslēdzas. Ievietojot jaunas baterijas
un atjaunojot pareizos jaudas līmeņus, sacietēšanas
gaisma atsāks darboties un atgriezīsies tajā pašāstāvoklī kā pirms jaudas zuduma. Cietēšanas gaisma
pašatjaunosies strāvas zuduma gadījumā.
Jaudas frekvences magnētiskajiem laukiem jābūt
tādiem, kas raksturīgi tipiskai atrašanās vietai tipiskā,dzīvojamā, mājas veselības aprūpes, tirdzniecības,slimnīcas vai militārā vidē.

1. piezīme: Cietēšanas gaisma nav aprīkota ar jebkādām pieslēgvietām vai pieejamajām I / O līnijām.

179

Vadlīnijas un Ražošanas deklarācija par elektromagnētisko noturību attiecībā uz nedzīvības atbalsta sistēmām

Vadlīnijas un Ražošanas deklarācija par ieteicamajiem attālumiem starp pārnēsājamām un mobilām RF sakaru iekārtām un VALO Bezvadu cietēšanas gaismu.

Cietēšanas gaisma ir paredzēta izmantošanai turpmāk norādītajā elektromagnētiskajā vidē. Klientam vai lietotājam būtu jānodrošina, ka tas tiek izmantots šādā vidē.

Cietēšanas gaisma ir paredzēta izmantošanai elektromagnētiskā vidē, kurā tiek kontrolēti radiācijas traucējumi. Cietēšanas gaismas lietotājs var palīdzēt novērst
elektromagnētiskos traucējumus, saglabājot minimālo attālumu starp pārnēsājamām un mobilām RF sakaru iekārtām (raidītājiem) un cietēšanas gaismu, kā norādīts
zemāk, atbilstoši sakaru iekārtu maksimālajai izejas jaudai.

IMMUNITĀTES tests

IEC 60601 testa
līmenisAtbilstības līmenis

Vadītspēja RF

3 Vrms3 VrmsIEC 61000-4-6

150 kHz līdz 80 MHz

150 kHz līdz 80 MHz

Radiālais RF

3V/m

3V/m

Elektromagnētiskās vides vadība
Pārnēsājamas un mobilās RF sakaru iekārtas nebūtu jāizmanto tuvu kādai no cietēšanas
gaismas daļām, ieskaitot kabeļus, nekā ieteicamais attālums, kas aprēķināts novienādojuma, ko piemēro raidītāja frekvencei.
Ieteicamais attālums:

80 MHz līdz 800 MHz

8800 MHz līdz 2,5 GHz
P ir raidītāja maksimālā izejas jauda vatos (W) atbilstoši raidītāja ražotājam un d ir
ieteicamais attālums metros (m).
Fiksēto RF raidītāju lauka stiprumiem, kas noteikti ar elektromagnētiskās vietas aptauju,
jābūt mazākiem par atbilstības līmeni katrā frekvenču diapazonā.

IEC 61000-4-3

80 MHz līdz 2,5 GHz

80 MHz līdz 2,5 GHz

Traucējumi var rasties iekārtu tuvumā, kas apzīmētas ar šādu simbolu:

Atdalīšanas attālums atkarībā no raidītāja frekvences(metri)Nominālā maksimālā izejas jauda
Raidītāja(P vatos)

150 kHz – 80 MHz

80 MHz – 800 MHz

800 MHz – 2,5 GHz

0,01

0,12 metri

0,035 metri

0,07 metri

0,1

0,37 m0,11 metri

0,22 metri

1

1,7 m

0,35 metri

0,7 metri

10

3,7 metri

1,11 metri

2,22 metri

100

11,7 metri

3,5 metri

7,0 metri

Cietēšanas gaisma ir pārbaudīta saskaņā ar IEC 60601-1-2: 2014 un nodota ar izstaroto lauka intensitāti 10 V / m no 80 MHz līdz 2,5 GHz. 3Vrms vērtība atbilst V1 unvērtība 10 V / m atbilst E1 iepriekšminētajās formulās.
Raidītājiem, kuru nominālā jauda nepārsniedz iepriekš minēto, ieteicamo atdalīšanas attālumu metros (m) var novērtēt, izmantojot vienādojumu, ko piemēro raidītāja
frekvencei, kur P ir raidītāja maksimālā jauda vatos ( W) pēc raidītāja ražotāja.
1. PIEZĪME 80 MHz un 800 MHz diapazonā piemēro augstāka frekvenču diapazona attālumu.
2. PIEZĪME Šīs vadlīnijas var nebūt piemērojamas visās situācijās. Elektromagnētisko izplatību ietekmē absorbcija un atstarošana no konstrukcijām, objektiem uncilvēkiem.

1. PIEZĪME 80 MHz un 800 MHz diapazonā piemēro augstāku frekvenču diapazonu.
2. PIEZĪME Šīs vadlīnijas var nebūt piemērojamas visās situācijās. Elektromagnētisko izplatību ietekmē absorbcija un atstarošana no konstrukcijām, objektiem uncilvēkiem.
Teorētiski nevar precīzi prognozēt lauka stiprumu no fiksētiem raidītājiem, piemēram, radio (šūnu / bezvadu) telefonu un sauszemes radiostaciju, amatieru
radio, AM un FM radio pārraides un televīzijas pārraides bāzes. Lai novērtētu elektromagnētisko vidi fiksēto RF raidītāju dēļ, ir jāapsver elektromagnētiskās vietas
apsekojums. Ja izmērītais lauka stiprums vietā, kurā tiek izmantota sacietēšanas gaisma, pārsniedz iepriekš noteikto RF atbilstības līmeni, jāpārbauda VALO bezvada
cietēšanas gaisma, lai pārbaudītu normālu darbību. Ja tiek novērota neparasta veiktspēja, var būt nepieciešami papildu pasākumi, piemēram, cietēšanas gaismaspārorientēšana vai pārvietošana.
b
frekvenču diapazonā no 150 kHz līdz 80 MHz lauka stiprumam jābūt mazākam par 3 V / m.

180

181

VALO™ GRANDLT

KIETINIMO LEMPA

Gaminio aprašymas

Dėl savo plačiajuosčio spektro „VALO Grand“ suprojektuota polimerizuoti visus šviesoje kietėjančius produktus 385-515 nm bangų ilgio diapazone pagal ISO 10650 reikalavimus.
„VALO Grand“ kietinimo lempa naudoja „Ultradent VALO" įkraunamus akumuliatorius ir akumuliatoriaus įkroviklį. Kietinimo lempa suprojektuota taip, kad būtų laikoma standartiniame
odontologinės įrangos laikiklyje arba gali būti montuojama nestandartiniu būdu naudojant rinkinyje esantį laikiklį.

Didžiosios VALO produkto komponentai:
•
•
•
•
•
•

Nuotrauka

Mygtukai ir lemputės

Kad nesugadintumėte įrangos, NEGALIMA kišti pirštų, instrumentų ar kitų daiktų į kietinimo lempos akumuliatoriaus skyrių.
Kad išvengtumėte įrangos sugadinimo, NEGALIMA valyti aukso kontaktų ar jokios baterijos skyriaus dalies. Jei kyla problemų, skambinkite „Ultradent“ klientų aptarnavimo tarnybai.
Siekiant išvengti kryžminio užteršimo pavojaus, barjerinės movos naudojamos tik vienam pacientui.
Kad sumažintumėte korozijos riziką, po naudojimo nuimkite apsauginę movą.
Kad sumažintumėte nepakankamai sukietėjusių dervų riziką, nenaudokite kietėjimo šviesos, jei pažeistas objektyvas.

Instrukcijos etapais
Paruošimas

1) Prieš naudodami kietinimo lempą, įkraukite akumuliatorius (žr. skyrių Akumuliatoriaus priežiūra).
2) Prieš kiekvieną naudojimą ant kietinimo lempos uždėkite naują apsauginę movą, kad lęšis kuo mažiau raukšlėtųsi ir būtų pasiekti geriausi rezultatai.

Pastaba:1 – „VALO Grand“ kietėjimo šviesa
4 – Ultradent VALO įkraunami akumuliatoriai
1 – „Ultradent VALO“ akumuliatoriaus įkroviklis su medicininiu 12 VDC kintamosios srovės maitinimo adapteriu
1 – VALO barjerinės movos pavyzdinis paketas
1 – „VALO Grand“ šviesos skydas
1 – Kietinimo lempos paviršiaus montavimo laikiklis su dvipuse lipnia juosta

Valdiklių apžvalga:

•
•
•
•
•

• Siekiant išvengti kryžminio užsikrėtimo ir kompozitinė medžiaga nepriliptų prie lęšio ir korpuso paviršiaus, kiekvieną kartą naudojant VALO kietinimo lempą reikia uždėti „Ultradent“
patvirtintą apsauginę movą. Apsauginės movos skirtos naudoti vienam pacientui..

Bevielės VALO šviesos skydas:

Belaidis VALO šviesos skydas yra ovalo formos, gali būti pasukamas maksimaliam naudojimui ir gali būti naudojamas su skaidria barjerine mova.

Naudojimas
PaaiškinimasĮjungimo mygtukas (-ai)(Viršuje arba apačioje)
Laiko lemputės
Žemas akumuliatoriaus įkrovos lygis –indikatoriusRežimo / būsenos lemputė
Laiko / režimo keitimo
mygtukas

1) Kiekvienas galios režimas naudojamas stomatologinių medžiagų su fotoiniciatoriais kietinimui. Rekomenduojamą kietėjimo laiką žr. Greitąjį režimo vadovą.
PASTABA: Kietinanti šviesa programuojama taip, kad iš standartinės galios į Didelės galios plius režimą būtų perjungta paeiliui per Ekstra galios režimą. Pavyzdžiui, norint pereiti iš
standartinio maitinimo režimo į Ekstra galios režimą, reikia pereiti į Aukštos galios režimą ir tada į Ekstra galios režimą.
2) Kietinimo šviesos lempa saugo paskutinį kartą naudojamą laiko intervalą ir režimą, ir pakeitus režimus arba išėmus baterijas, automatiškai bus grąžintas šis režimas.

DarbasKIETINIMO REŽIMAS: Standartinis galios režimasLAIKO INTERVALAI: 5, 10, 15, ir20 sekundžių.
• Kietinimo šviesos lempa automatiškai įsijungia į šį režimą, kai jis įjungiamas IŠ PRADŽIŲ. Režimas / būsenos lemputė užsidegs žalia spalva ir užsidegs keturios žalios šviesos laikolemputės, nurodydamos standartinį galios režimą.
• Norėdami pakeisti laiko intervalus, greitai spauskite laiko / režimo mygtuką.
• Norėdami kietinti, paspauskite įjungimo mygtuką. Norint sustabdyti kietinimą prieš pasibaigiant laiko intervalui, dar kartą paspauskite įjungimo mygtuką.

KIETINIMO REŽIMAS: Didelės galios plius režimas

Apšvietimo šaltinis šviesa aktyvinamoms stomatologinėms restauravimo medžiagoms ir adhezivams kietinti.

LAIKO INTERVALAI: 1, 2, 3, ir4 sekundžių.
• Standartinės galios režimu 2 sekundes palaikykite nuspaudę laiko / režimo keitimo mygtuką ir atleiskite. Režimas / būsenos lemputė bus oranžinė, o keturios žalios laiko lemputės
užsidegs ir mirksės, nurodydamos didelės galios režimą.
• Norėdami pakeisti laiko intervalus, greitai spauskite laiko / režimo mygtuką.
• Norėdami kietinti, paspauskite bet kurį įjungimo mygtuką. Norėdami sustabdyti kietėjimą prieš pasibaigiant laiko intervalui, dar kartą paspauskite bet kurį mygtuką.
• Norėdami grįžti į standartinės galios režimą, palaikykite nuspaudę laiko / režimo keitimo mygtuką 2 sekundes ir atleiskite. Tokiu būdu pereisite į Ekstra galios režimą. 2 sekundes
palaikykite paspaudę ir atleiskite. Režimo / būsenos lemputė užsidegs žaliai ir užsidegs keturios žalios lemputės, nurodančios standartinį režimą.

Įspėjimai ir atsargumo priemonėsKIETINIMO REŽIMAS: Ekstra galios režimas

(Bottom)(Top)

Prieš naudojimą atidžiai perskaitykite ir supraskite visų aprašytų gaminių visas instrukcijas ir SDL pateikiamą informaciją.

Naudojimo / numatomo tikslo indikacijos

2 rizikos grupė
ATSARGIAI šis produktas skleidžia UV. Poveikis gali sukelti akių ar odos dirginimą. Naudokite atitinkamą ekranavimą.
ATSARGIAI Šis produktas skleidžia potencialiai pavojingą optinę radiaciją. Nežiūrėkite į operacinę lempą. Gali būti kenksminga akims.
• NEŽIŪRĖKITE tiesiai į šviesos šaltinį. Kai naudojama kietinimo lempa, pacientas, gydytojas ir asistentai visada turi būti užsidėję gintaro spalvos apsauginius UV akinius.
• Siekiant išvengti elektros smūgio pavojaus, draudžiama bet kokia šios įrangos modifikacija. Naudoti tik pridedamą „Ultradent VALO” maitinimo šaltinį ir kištukinius adapterius. Jei šiossudedamosios dalys pažeistos, nenaudokite ir skambinkite į "Ultradent" klientų aptarnavimo tarnybą, kad užsakytumėte pakeitimą.
• Nešiojama radijo dažnio ryšio įranga gali pabloginti našumą, jei ji naudojama arčiau nei 30 cm (12 col.).
• Naudokite tik patvirtintus priedus, laidus ir maitinimo šaltinius, kad išvengtumėte netinkamo veikimo, padidėjusio elektromagnetinio spinduliavimo arba sumažėjusio
elektromagnetinio atsparumo (žr. skyrių „Elektromagnetinė emisija“).
• Kad išvengtumėte elektros gaisro pavojaus, susijusį su akumuliatorių tvarkymu:
⸰ NEMAIŠYTI įkraunamų akumuliatorių su neįkraunamomis baterijomis ar kitomis baterijų rūšimis.
⸰ NEGALIMA įkrauti neįkraunamų baterijų. Neįkraunamų akumuliatorių įkrovimas gali sukelti sužalojimų arba sugadinti turtą.
⸰ NENAUDOKITE sterilizatorių ir nepurkškite akumuliatorių, akumuliatoriaus kontaktų, įkroviklio ar kintamosios srovės maitinimo adapterio jokiu skysčiu. Jei ant akumuliatoriausįkroviklio kontaktų atsiranda korozija, paskambinkite „Ultradent“ klientų aptarnavimo tarnybai ir užsakykite pakeitimą.
⸰ NEKRAUKITE akumuliatorių šalia degių medžiagų.
⸰ NELAIKYKITE įkroviklio klinikos operacinėje.
• Kad išvengtumėte sužeidimo pavojaus, NENAUDOKITE baterijų, kurios yra paveiktos korozijos (surūdijusios), įlenktos, skleidžia kvapą ar skysčius, kurių pakuotė yra suplėšyta arba josnėra, arba yra kitaip pažeistos. Norėdami užsisakyti atsarginius akumuliatorius, kreipkitės į „Ultradent" klientų aptarnavimo tarnybą.
• Naudokite tik rekomenduojamus akumuliatorius. Alternatyvūs akumuliatoriai gali sukelti veikimo sutrikimų.
• Siekiant išvengti šiluminio dirginimo ar nudeginimo, venkite iš eilės einančių kietinimo ciklų, o taip pat minkštieji burnos audiniai neturėtų būti veikiami iš arti ilgiau nei 10 sek. bet
kuriuo režimu. Jei reikia ilgesnio kietėjimo laiko, naudokite kelis kietėjimo ciklus su poilsio laikotarpiais tarp ciklų arba naudokite dvigubo kietėjimo produktą, kad išvengtumėte
minkštųjų audinių įkaitimo.
• Būkite atsargūs gydydami pacientus, kuriems pasireiškia neigiamos fotobiologinės reakcijos arba jie yra šiems procesams jautrūs, pacientus, kurie gydomi chemoterapija arpacientus,
kurie gydomi šviesai jautriais vaistais.
• Šis įrenginys gali būti jautrus stipriems magnetiniams ar statiniams elektriniams laukams, nes jie gali sutrikdyti programą. Jei įtariate, kad taip atsitiko, nedelsdami išjunkite įrenginį iš
elektros lizdo ir vėl jį prijunkite prie elektros lizdo.
• NEVALYKITE kietinimo lempos šarminiais ar abrazyviniais valikliais, sterilizatoriuje, nenardinkite į jokią ultragarsinę vonia, dezinfekcijos medžiagą, valymo tirpalą ar skystį. Jei
nesilaikysite pridedamų naudojimo instrukcijų, kietinimo lempa gali tapti nebetinkama naudojimui.
182

LAIKO INTERVALAS: Tik 3 sekundės (Pastaba: Ekstra galios režimas kiekvieno kietinimo ciklo pabaigoje turi 2 sekundžių saugos atidėjimą, kad būtų sumažintas nenutrūkstamas
kaitinimas kietinimo metu. Atidėjimo pabaigoje pyptelėjimas nurodo, kad įrenginys yra paruoštas naudoti toliau.
• Standartinės galios režimu 2 sekundes spauskite Laiko / režimo pakeitimo mygtuką, atleiskite, paspauskite ir vėl palaikykite 2 sekundes, tada atleiskite. Režimas / būsenos lemputė
švies oranžine spalva ir mirksės, o trys žalios laiko lemputės užsidegs ir mirksės, nurodančios Ekstra galios režimą.
• Norėdami kietinti, paspauskite įjungimo mygtuką. Norėdami sustabdyti kietinimą prieš pasibaigiant laiko intervalui, dar kartą paspauskite maitinimo mygtuką.
• Norėdami grįžti į standartinės galios režimą, laikykite nuspaudę laiko / režimo mygtuką 2 sekundes ir atleiskite. Režimo / būsenos lemputė užsidegs žaliai ir švies žalios laikolemputės, nurodančios standartinės galios režimą.

Budėjimo režimas: Kietinimo šviesos lempa perjungiama į budėjimo režimą po 60 sekundžių neveiklumo, kuris nurodomas lėtai mirksinčia režimo / būsenos lempute. Įrenginį pakėlus
arba palietus, kietinimo šviesos lempos budėjimo režimas automatiškai išsijungs ir bus grąžintas paskutinis naudotas nustatymas. Kad akumuliatoriaus tarnavimo laikas būtų kuo
ilgesnis, kai nenaudojate, kietinimo šviesos lempos nejudinkite.

Valymas1) Kiekvieno paciento panaudotas barjerines movas išmeskite į standartines atliekas.
2) Žr. skyrių Apdirbimas.

Montavimo laikiklio instrukcijos

1) Laikiklis turi būti montuojamas ant lygaus, tepalais nesutepto paviršiaus.
2) Paviršių nuvalykite įtrindami alkoholiu.
3) Nulupkite nuo laikiklio lipnią juostą.
4) Nustatykite laikiklio padėtį taip taip, kad kietinimo šviesos lempą ištraukus būtų nukreipta aukštyn. Tvirtai įspauskite į vietą.

183

Greito režimo vadovas:
Standartinė galia
1000 mW/cm2

RežimasDidelė galia plius
1600 mW/cm2

Ekstra galia
3200 mW/cm2

Įjungimo mygtukasRežimo / laiko nustatymo LED

Laiko mygtukai

Laiko parinktys

5s

10s

15s

20s

1s

2s

3s

4s

Tik 3 sekNorėdami pakeisti laiką

Paspauskite ir greitai atleiskite laiko mygtuką, jei norite pereiti per laiko parinktis.

Norėdami pakeisti režimus

Paspauskite ir laikykite nuspaudę laiko mygtuką 2 sekundes ir atleiskite. Kietinimo lempa pereis į kitą režimą.

Paaiškinimai

Tolygiai šviečiantys LED

Mirksintys LED

Kietinimo režimas

Galia * (mW)Spinduliavimas*(mW/cm2)

Bendras ekspozicijos laikas (sek.)

Energija**(Džauliai)

Standartinis

970

900

5/10/15/20

4,8 J / 9,7 J / 14,5
J/19,4 J

Didelė galia plius

1,620

1,500

1/2/3/4

1,6 J / 3,2 J / 4,8 J / 6,5 J

Ekstra

2,260

2,100

3

6,8 J

*Nominalioji galia 2 mm atstumu nuo lęšio galo iki viršutinio kompozito paviršiaus.
**Bendros energijos per ciklą (džaulių) dalyje pateiktos vertės suapvalintos iki artimiausios dešimtosios dalies.

Įspėjimas dėl galios lygioĮspėjimas apie temperatūrą

Įspėjimas apie techninį aptarnavimą

LED įspėjimas

Pakeiskite akumuliatorius

Leiskite atvėsti

Dėl remonto skambinkite Klientų
aptarnavimo tarnybai

Dėl remonto skambinkite Klientų
aptarnavimo tarnybai• Nėra garso• Mirksi kas 2 sekundes• Leidžia atlikti veiksmą

• Nuolatiniai 3 pyptelėjimai• Viena greita mirktelėjimas• Draudžiami veiksmai• Žemas akumuliatoriaus įkrovos
lygis: lėtai mirksi• Išjungti: 3 pyptelėjimai, mirksi• Neleidžia atlikti veiksmo• 3 pyptelėjimai• Lėtai mirksi• Neleidžia atlikti veiksmo

Kaip įkrauti akumuliatorius:
1) Įkroviklį ir akumuliatorius padėkite tokioje vietoje, kur jie nesiliečia su skysčiais ar dezinfekavimo purškiamaisiais skysčiais.
2) Įdėkite akumuliatorius į įkroviklį teigiamą (+) galą nukreipę link įkroviklio indikatoriaus lempučių.
3) Žalia šviesa nurodo, kad baterijos yra paruoštos naudoti.
4) Akumuliatorių įkrovimas užtruks 1–3 valandas. Palikite akumuliatorius įkroviklyje, kol jie bus paruošti naudoti.
PASTABA: Jei įkroviklio raudona lemputė neužsidega žaliai, kai akumuliatoriai įkraunami ilgiau nei tris valandas, akumuliatorius gali būti sugedęs (pažeistas) ir jo negalima įkrauti.
Išbandykite naują akumuliatorių arba paskambinkite „Ultradent“ klientų aptarnavimo tarnybai, kad užsisakytumėte naują įkraunamų akumuliatorių rinkinį.
Kaip pakeisti / įdėti akumuliatorius:
1) Galinį dangtelį nuimkite, sukdami ketvirtį apsisukimo prieš laikrodžio rodyklę.
2) Išimkite akumuliatorius.
3) Pirmiausia įdėkite naujų baterijų teigiamą (+) pusę.
4) Galinį dangtelį sulygiavę įdėkite atgal ir švelniai stumkite sukdami pagal laikrodžio rodyklę. Dangtelis spragtelės, kai visiškai įsitvirtins.
5) Prietaisas paruoštas naudojimui.
Žemas akumuliatorių (baterijų) įkrovos lygis: kai mirksi raudona žemo akumuliatoriaus įkrovos lygio indikatoriaus lemputė, kietinimo lempos signalas informuoja naudotoją, kadlaikas keisti akumuliatorius. Jei akumuliatoriaus įkrovos lygis tampa pernelyg žemas, pasigirs 3 pyptelėjimai ir su kietinimo lempa nebus galima atlikti jokių tolesnių operacijų, kol
akumuliatoriai nebus įkrauti arba nebus įdėtos naujos baterijos. (Žr. greito įspėjimo vadovą).

Greitasis kietinimo vadovas:

Greitojo įspėjimo vadovas:

Vartotojo atliekama priežiūra
1) Naudokite barjerinę movą, kad dantų kompozitas nepriliptų prie lęšio paviršiaus. Jei reikia, naudokite plastikinį arba nerūdijančio plieno dantų instrumentą, kad atsargiai
pašalintumėte prilipusį kompozitą. Nenaudokite įrankių, kurie gali pažeisti lęšį.
2) Šviesos matuokliai labai skiriasi ir yra skirti konkretiems šviesos kreipimo antgaliams ir lęšiams. „Ultradent“ rekomenduoja reguliariai tikrinti išvestį standartinio maitinimo režimu.
PASTABA: tikroji skaitmeninė išvestis bus iškreipta dėl įprastų šviesos matuoklių netikslumo ir pasirinktinio kietinimo lempos LED paketo.
Akumuliatoriaus priežiūra
Akumuliatorių įkrovimas ir keitimas
Kietinimo lempa tiekiama su 4 įkraunamais ličio geležies fosfato akumuliatoriais. „Ultradent“ rekomenduoja naudoti tik VALO įkraunamus akumuliatorius.

Įkrovimo laikas ir akumuliatoriaus tarnavimo laikas: kietinimo lempos visiškai įkrauto akumuliatoriaus veikimo laikas priklauso nuo režimo / laiko intervalo, akumuliatoriaus tipo,
naudojimo intensyvumo ir LED efektyvumo.
• Rekomenduojamas įkrovimo intervalas: įkraukite pagal poreikį arba kai užsidega išsikrovusio akumuliatoriaus indikatorius. Veikimo trukmė priklauso nuo naudojimo.
• Papildomi akumuliatoriai: kietinimo lempoje yra 4 įkraunami akumuliatoriai.
Gamintojo remontas
1) Remontą gali atlikti tik įgalioti techninės priežiūros personalas. „Ultradent" teikia techninės priežiūros personalui dokumentus, reikalingus remontui atlikti.
2) Siųsdami prietaisus remontui, aptarnavimui ar kalibravimui, visada išimkite akumuliatorius iš kietinimo lemputės ir įkroviklio. Akumuliatorius, įkroviklį, adapterį ir kietinimo lempą
atskirai supakuokite į grąžinimo dėžutę.
3) Akumuliatorius siųskite laikydamiesi vietinių taisyklių.
Garantija„Ultradent Products, Inc.“ („Ultradent“) garantuoja, kad šis gaminys galioja 5 metus nuo įsigijimo datos, kai jis naudojamas pagal su gaminiu pateiktas naudojimo instrukcijas, (i) visais
esminiais atžvilgiais atitiks „Ultradent“ prie gaminio pridėtuose dokumentuose nurodytas specifikacijas; ir (ii) neturės medžiagų ir gamybos defektų.
Ši ribota garantija neperleidžiama ir taikoma tik pirminiam pirkėjui, ji netaikoma vėlesniems gaminio savininkams. Ši ribota garantija netaikoma jokiems kitiems priedų komponentams,
pvz., akumuliatoriams, įkrovikliams, adapteriams ar adaptyviniams lęšiams, tačiau jais neapsiriboja. Ši ribota garantija negalioja, jei gaminys sugenda arba yra sugadintas dėl aplaidumo,
piktnaudžiavimo, netinkamo naudojimo, nelaimingo atsitikimo, modifikavimo, klastojimo, pakeitimo arba galiojančių naudojimo instrukcijų nesilaikymo. Tik pavyzdžiui, nukritusiam irpažeistam gaminiui ši garantija netaikoma. Norint pasinaudoti šia ribota garantija, kartu su gaminiu su defektais „Ultradent“ reikia pateikti pirkimo įrodymą (pvz., pardavimo kvitą arpanašius dokumentus).
Defektinis gaminys, atitinkantis čia nustatytas garantijos sąlygas, „Ultradent“ nuožiūra bus pataisytas arba pakeistas. „Ultradent“ atsakomybė už gaminį jokiu būdu neviršija pirkėjo
sumokėtos pirkimo kainos. Jokiomis aplinkybėmis „Ultradent“ neatsako už bet kokią netiesioginę, atsitiktinę, nenumatytą, nenumatytą, specialią ar netiesioginę žalą, atsiradusią dėl šio
gaminio naudojimo ar susijusią su juo.

Apdirbimas

Šis gaminys skirtas naudoti su barjerine mova. Apie priežiūrą ir bendrą valymą žr. aukščiau.

Laikymas ir šalinimas

Jei kietinimo lempa saugoma ilgesnį nei 2 savaičių laikotarpį arba supakuojama kelionei, visada išimkite akumuliatorius. Įrenginyje ilgam laikui palikus nekraunamus akumuliatorius jie
gali pati nebetinkami arba nebeįkraunami. Negalima laikyti akumuliatorių aukštesnėje nei 40 °C (104 °F) temperatūroje arba tiesioginėje saulės šviesoje.

Priežiūra

Bendras kietinimo lempos valymas
Po kiekvieno naudojimo sudrėkinkite marlę arba minkštą šluostę patvirtinta paviršiaus dezinfekavimo priemone ir nuvalykite paviršių bei lęšius. Neleistinos valymo priemonės gali
sugadinti kietinimo lempą.
TINKAMI VALIKLIAI:• 70 % izopropilo alkoholis• 70 % etanolis

Bendras šviesos skydo valymas:
Valykite belaidį VALO šviesos skydą naudodami bet kokią paviršiaus dezinfekavimo priemonę. DRAUDŽIAMA autoklavuoti.
Remontas
184

Kietinimo lempos laikymas ir transportavimas:• Temperatūros intervalas: nuo + 10 °C iki + 40 °C (+ 50 °F iki + 104 °F)• Santykinė drėgmė: nuo 10 % iki 95 %• Aplinkos slėgis: 500 hPa – 1060 hPa
Išmesdami elektronines atliekas (t. y. kietinimo lempas, įkroviklius, akumuliatorius ir maitinimo šaltinius), vadovaukitės vietinėmis atliekų šalinimo ir perdirbimo gairėmis.

Techniniai aspektai
Priedai185

Elementas

CE informacija

Techninė informacija / duomenys

Efektyvios kompozito kietinimo bangų ilgių juostos:

Belaidis VALO šviesos skydas
EFEKTYVIOS KOMPOZICINĖS KIETĖJIMO BANGELĖS JUOSTOS(Naudojamas bangos ilgio diapazonas: 385–515 nm)
SPEKTRALINĖ SPINDULIO GALIA (mW/nm)

VALO įkroviklis

VALO akumuliatoriai

Veiksmingasžemesnisbangos ilgio juosta
395-415 nm

Efektyvi viršutinė
bangos ilgio juosta
440-480 nm

EkstraDidelė galia
Standartinis

BANGOS ANGLIJA (nm)

Savybė

Informacija / specifikacija

Lęšis

Skersmuo 11,7 mm

Bangos ilgio
diapazonas• Naudojamas bangos ilgio diapazonas: 385 - 515 nm• Piko bangos ilgiai: 395 - 415 nm ir 440 - 480 nm
Nominalaus spindulinio šviesio palyginimo diagrama
Matavimo
priemonė

Šviesos intensyvumo lentelė

186‡ Gigahertz“ spektro analizatoriusŠviesio

bendra galia

Matuoklio
diafragma

15 mm

15 mm

Standartinė galia(±10%)

900 mW/cm²

970 mW

Didelė galia plius(±10%)

1500 mW/cm²

1615 mW

Ekstra galia (±10%)2100 mW/cm²

Spindulinis šviesis skirsis priklausomai nuo
prietaiso galimybių, matavimo metodo iršviesos išdėstymo.
‡ Išmatavus „Gigahertz-Optik“ spektro
analizatoriumi, spindulinis šviesis atitinka ISO
10650 reikalavimus.
.

2260 mW

Didžioji VALO
kietinimo lempa

Normos: IEC 60601-1 (Sauga), IEC 60601-1-2 (EMC)

Svoris:• su baterijomis: 190 gramų (6 unc.)• be baterijų: 150 gramų (5 unc.)
Matmenys: (203 x 32,5 x 27) mm, (8 x 1,28 x 1,06) colių

Įkroviklio
maitinimasIšėjimas – 12 VDC esant 500 mAĮėjimas – 100 VAC–240VAC„Ultradent“ P/N 5930 VALO įkroviklio maitinimas su
tarptautiniais kištukais

Normos: IEC 60601-1 (Sauga)
Laido ilgis – 1,8 m (6 pėdos)VALO įkroviklio maitinimo šaltinis yra medicininės klasės II klasės maitinimošaltinis su izoliacija nuo elektros tinklo maitinimo187Įvairi informacija

Savybė

Informacija / specifikacija

Nurodymai ir gamintojo deklaracija dėl elektromagnetinių emisijų

VALO įkroviklis

VALO 3,6 VDC ličio geležies fosfato išmanusis akumuliatorių įkroviklis:• Automatinis išjungimas visiškai įkrovus• Automatinis sugedusių akumuliatorių aptikimas• Apsaugos: šilumos, perkrovos, trumpo jungimo, atvirkštinio poliariškumo
⸰ raudonas LED – įkraunama⸰ žalias LED – tuščia arba visiškai įkrauta
⸰ LED išjungtas – trumpasis jungimas• Įkrovimo laikas: 1 – 3 val.
Normos: CE, WEEE

Emisijų bandymas

Atitiktis

Elektromagnetinė aplinka

RF spinduliuotėCISPR 11

1 grupė

Kietinimo lempa naudoja akumuliatorius ir jai EMI, RF ar viršįtampių slopinimas įtakos neturi.

RF emisija
CISPR 11

B klasė

VALO akumuliatoriaiĮkraunami: saugi ličio geležies fosfato (LiFePO4) chemija RCR123A• Darbinė įtampa: 3,2 VDC, 400 mAh (reikalinga vatvalandė)Normos: UL 62133-2, CE, RoHS, WEEE
Skaičiuojama, kad elementai gali atlikti >2000 įkrovimo ir iškrovimo ciklų, kol pasiekia 70 % savo talpos. Akumuliatorius patartina keisti
kas 3-5 metus, tačiau jų tarnavimo laikas gali skirtis priklausomai nuo naudojimo. Jei reikia, pakeiskite.

Kietinimo lempa naudoja elektros ir elektromagnetinę energiją tik savo vidaus funkcijoms.
Todėl bet kokios radijo dažnių spinduliuotės yra labai mažos ir netrukdo veikti netoliese esančiai
elektroninei įrangai.

Harmonikų emisijos
IEC 61000-3-2

NėraHarmonikų emisijos ir įtampos svyravimo bandymai netaikomi kietinimo lempai, nes ji maitinama
akumuliatoriaus energija.

Naudojimo sąlygos

Temperatūros intervalas: +10 °C – +32°C (+50 °F – +90 °F)
Santykinė drėgmė: nuo 10 % iki 95 %Aplinkos slėgis: 700 hPa – 1060 hPaĮtampos svyravimai / mirgėjimas NėraIEC 61000-3-3

Kietinimo lempa tinkama naudoti visose įstaigose, įskaitant vidaus įmones ir tiesiogiai prijungtus
prie viešo žemos įtampos maitinimo tinklo, tiekiančiam elektrą pastatuose buitiniam naudojimui.

Darbo ciklas

Kietinimo šviesa skirta trumpalaikiam darbui. Maksimali aplinkos temperatūra (32 ° C) 1 minutės ĮJUNGIMAS gretimi ciklai, 30 minučiųIŠJUNGIMAS (išjungimo laikotarpis).

Nurodymai ir gamintojo deklaracija dėl elektromagnetinio suderinamumo
Kietinimo lempa skirta naudoti žemiau nurodytoje elektromagnetinėje aplinkoje. Klientas ar naudotojas privalo užtikrinti, kad ji būtų naudojama tokioje aplinkoje.

Trikčių šalinimas
Jei toliau siūlomi sprendimai nepadeda pašalinti trikties, skambinkite „Ultradent“ telefonu 800.552.5512. Už JAV ribų skambinkite savo „Ultradent“ platintojui arba
odontologiniam pardavėjui.
Problema

188

Kietinimo lempa skirta naudoti žemiau nurodytoje elektromagnetinėje aplinkoje. Klientas ar naudotojas privalo užtikrinti, kad ji būtų naudojama tokioje aplinkoje.

Galimi sprendimaiNeįmanoma įjungti lempos

1) Paspauskite laiko / režimo arba įjungimo mygtuką, jei norite išjungti energijos taupymo režimą.
2) Patikrinkite, ar akumuliatoriaus įkrovos būseną nurodantis žemos akumuliatoriaus įkrovos indikatorius šviečia raudonai.
3) Patikrinkite, ar į prietaisą teisingai įdėtos naujos baterijos.
4) Jei mirksi raudoni ir geltoni įspėjamieji LED, kietėjimo kietinimo lempa pasiekė vidinės temperatūros apsauginę ribą.
Leiskite 10 minučių kietinimo lempai atvėsti arba naudokite vėsų drėgną rankšluostį, kad prietaisas būtų galima greitaiatvėsinti.
5) Jei raudonas įspėjamasis LED mirksi ir nuolat pypsi, skambinkite „Ultradent“ klientų aptarnavimo tarnybai.

Lempa neišbūna įsijungusi reikiamą
laiką1) Patikrinkite režimo ir laiko nustatymo lemputes, ar laiko įvestis nustatyta teisingai.
2) Patikrinkite žemo akumuliatoriaus įkrovos lygio indikatorių ar akumuliatorius neišsikrovęs.
3) Patikrinkite, ar į prietaisą tinkamai įdėta naujų baterijų.

Šviesa netinkamai kietina dervas

1) Patikrinkite lęšį, ar ant jo nelikę kietintos dervos / kompozitų likučių.
2) Naudodami tinkamą gintaro UV akių apsaugą, patikrinkite, ar veikia LED lemputės.
3) Šviesos matuokliu patikrinkite galios lygį. Jei naudojate šviesos skaitiklį, „Ultradent“ rekomenduoja tikrinti kietinimolempą standartinės galios režimu.
PASTABA: Tikroji skaitmeninė išvestis bus iškreipta dėl įprastų šviesos matuoklių netikslumo ir pasirinktinio kietinimo lempos
LED paketo. Šviesos matuokliai labai skiriasi ir yra skirti konkretiems šviesos kreipimo antgaliams ir lęšiams.
4) Patikrinkite kietinamos dervos galiojimo datą.
5) Patikrinkite, ar laikomasi tinkamos procedūros pagal gamintojo rekomendacijas.

Neįmanoma įkrauti akumuliatorių

1) Įsitikinkite, kad akumuliatoriai į įkroviklį įdėti tinkamai ir palaukite 1–3 valandas, kad įkrautumėte akumuliatorius.
2) Jei įkroviklio raudona lemputė neužsidega žaliai, skambinkite „Ultradent“ klientų aptarnavimo tarnybai, kad užsakytumėte
atsargines baterijas ir (arba) įkroviklį.
3) Jei nematomos nei žalia, nei raudona įkroviklio lemputės, skambinkite „Ultradent“ klientų aptarnavimo tarnybai, kadužsakytumėte ar pakeistumėte įkroviklį ir (arba) kintamosios srovės adapterį.

Įkroviklis neįkrauna akumuliatorių

1) Įsitikinkite, kad įkroviklis yra prijungtas ir kintamosios srovės adapteris prijungtas prie veikiančio maitinimo lizdo.
2) Jei nematomos žalia arba raudona įkroviklio lemputės, skambinkite „Ultradent“ klientų aptarnavimo tarnybai, kadužsakytumėte naują įkroviklį ir (arba) kintamosios srovės adapterį.

Negalima pakeisti režimo ar laikointervalų

1) Abu laiko / režimo ir įjungimo mygtukus laikykite nuspaudę, kol pasigirs pyptelėjimų serija, reiškianti, kad kietinimo
lempa atrakinta.

SUDERINAMUMO testas

IEC 60601 bandymo lygis

Atitikties lygis

Elektromagnetinės aplinkos nurodymai

Elektrostatinė iškrova (ESD)±8 kV kontaktas±15 kV oro±8 kV kontaktas±15 kV oro

Fizinė aplinka turėtų apsiriboti:
1) IP kodas: IP20
2) Nemerkti į skystį.
3) Nenaudoti, kai aplinkoje yra degių dujų.
Įrenginys yra ne APG ir ne AP.
4) Sandėliavimo drėgmės intervalas: 10 % – 95 %5) Sandėliavimo temperatūrų intervalas: 10°C – 40 °C

Trumpalaikė pasikartojanti srovė
/ pertrūkiai
IEC 61000-4-4±2 kV maitinimo linijoms±1 kV įvesties / išvesties linijoms±2 kV maitinimo linijoms
1 pastaba: Kietinimo lempa neturiįvesties / išvesties prievadų

Kietinimo lempa maitinama akumuliatoriais ir jos
negalima prijungti prie kintamos srovės tinklo
maitinimo.

Viršįtampis
IEC 61000-4-5Viršįtampis
IEC 61000-4-5±1 kV iš linijos į liniją
±2 kV į žemę iš linijosĮtampos kritimai, trumpieji
jungimai, trikdžiai ir svyravimai
elektros maitinimo tinkluose<5 % U(>95 % U kritimas 0,5 ciklo)<5 % U(>95 % U kritimas 0,5 ciklo)

Kadangi kietinimo lempa yra maitinama
akumuliatoriais, ji neturi trumpalaikio elektros
laidumo, viršįtampio, įtampos kritimo, trumpų
jungimų, pertrūkių ar kintamosios srovės tinklo
maitinimo svyravimų.

40 % U(60 % U kritimas 5 ciklams)

40% U(60% U kritimas 5 ciklams)

70 % U(30 % U kritimas 25 ciklams)

70% U(30% U kritimas 25 ciklams)<5 % U(>95 % U kritimas 5 sek.)<5 % U(>95 % U kritimas 5 sek.)
2 pastaba: Savaime atsinaujina

30 A/m30 A/mIEC 61000-4-2

IEC 61000-4-11

Maitinimo dažnis
(50/60 Hz) magnetinis laukas
IEC 61000-4-8

Pridedamas akumuliatorių įkroviklis gali būti veikiamas
aukščiau išvardytų veiksnių, tačiau jis yra atskirtas irneįtakoja belaidės VALO kietinimo lempos darbo.
Jei kietinimo lempos akumuliatoriaus įtampa nukrenta
iki 4 VDC, įrenginys neveiks. Belaidė VALO kietinimo
lempa išsijungs. Įdėjus naujus akumuliatorius ir
atkuriant tinkamus galios lygius, vėl pradėjusi veikti
kietinimo lempa ir grįš į tą pačią būseną, kurioje buvo
prieš nutrūkstant maitinimui. Nutrūkus maitinimui,kietinimo lempa atsigauna savaime.
Maitinimo dažnio magnetiniai laukai turi būti būdingi
tipinei vietai tipinėje, gyvenamojoje, sveikatos
priežiūros namuose, komercinėjei, ligoninės arkarinėje aplinkoje.

1 pastaba: Kietinimo šveisa neturi jokių prievadų ar prieinamų įvesties / išvestis linijų.

189

Nurodymai ir gamintojo deklaracija dėl ne gyvybės palaikymo sistemų elektromagnetinio suderinamumo

Nurodymai ir gamintojo deklaracija dėl rekomenduojamų atstumų tarp nešiojamų ir mobiliųjų radijo ryšio įrenginių ir belaidės VALO kietinimo lempos

Kietinimo lempa skirta naudoti žemiau nurodytoje elektromagnetinėje aplinkoje. Klientas ar naudotojas privalo užtikrinti, kad ji būtų naudojama tokioje aplinkoje.

Kietinimo lempa skirta naudoti elektromagnetinėje aplinkoje, kurioje spinduliuojami radijo dažnio trikdžiai valdomi. Kietinimo lempos naudotojas gali padėti užkirsti
kelią elektromagnetiniams trikdžiams išlaikydamas minimalų atstumą tarp nešiojamųjų ir mobiliųjų radijo ryšio įrenginių (siųstuvų) ir kietinimo lempos, kaip nurodyta
toliau, atsižvelgiant į maksimalią komunikacijos įrangos išėjimo galią.

SUDERINAMUMO
testasIEC 60601 bandymo
lygis

Atitikties lygis

RF laidumas

3 Vrms3 VrmsIEC 61000-4-6

150 kHz – 80 MHz

150 kHz – 80 MHz

Spinduliuojamas
radijo bangų dažnis

3 V/m

3 V/m

Elektromagnetinės aplinkos nurodymai
Nešiojamoji ir judrioji radijo ryšio įranga turėtų būti naudojama šalia bet kokios kietinimo
lempos dalies, įskaitant kabelius, ne arčiau, nei rekomenduojamas atskyrimo atstumas,
apskaičiuotas pagal siųstuvo dažniui taikomą lygtį.
Rekomenduojamas atskyrimo atstumas:

80 MHz – 800 MHz

800 MHz – 2,5 GHz
P yra siųstuvo didžiausia išėjimo galia vatais (W) pagal siųstuvo gamintoją ir d yra
rekomenduojamas atstumas metrais (m).
Fiksuoto radijo dažnio siųstuvų lauko stiprumai, nustatyti pagal elektromagnetinį vietos
tyrimąa, , turėtų būti mažesni už atitikties lygį kiekviename dažnių diapazoneb.

IEC 61000-4-3

80 MHz – 2,5 GHz

80 MHz – 2,5 GHz

Trikdžiai gali atsirasti šalia įrangos, pažymėtos šiuo simboliu:

Atskyrimo atstumas pagal siųstuvo dažnį
(metrai)Nominali siųstuvo maksimali išėjimo
galia(P vatais)

150 kHz – 80 MHz

80 MHz – 800 MHz

800 MHz – 2,5 GHz

0,01

0,12 m0,035 m0,07 m0,10,37 m0,11 m0,22 m1

1,7 m

0,35 m

0,7 m

10

3,7 m

1,11 m2,22 m100

11,7 m

3,5 m

7,0 m

Kietinimo lempai buvo atlikti bandymai pagal IEC 60601-1-2: 2014 ir patvirtinta įvertinus spinduliuotės lauko stiprumą 10 V/m nuo 80 MHz iki 2,5 GHz. 3 Vrms vertė
atitinka V1, o vertė 10 V/m atitinka E1 aukščiau nurodytose formulėse.
Siųstuvams, kurių maksimali išėjimo galia nenurodyta pirmiau, rekomenduojamas atskyrimo atstumas d metrais (m) gali būti įvertintas naudojant siųstuvo dažniui
taikomą lygtį, kur P yra siųstuvo didžiausia išėjimo galia vatais (W) pagal siųstuvo gamintoją.
1 PASTABA. Esant 80 MHz ir 800 MHz dažnių diapazonui, taikomas didesnio dažnio diapazono atskyrimo atstumas.
2 PASTABA Šios gairės gali būti netinkamos visose situacijose. Elektromagnetinį sklidimą veikia konstrukcijų, objektų ir žmonių sugertis ir atspindėjimas.

1 PASTABA 80 MHz ir 800 MHz dažnių diapazone taikomas didesnis dažnių diapazonas.
2 PASTABA Šios gairės gali būti netinkamos visose situacijose. Elektromagnetinį sklidimą veikia konstrukcijų, objektų ir žmonių sugertis ir atspindėjimas.
Teoriškai negalima tiksliai nustatyti prognozuojamų siųstuvų, pvz., radijo (mobiliųjų / belaidžių) telefonų ir sausumos judriojo radijo, mėgėjų radijo, AM ir FM radijotransliacijų ir televizijos transliacijų lauko stiprio. Norint įvertinti elektromagnetinę aplinką, įvertinant fiksuotus radijo siųstuvus, reikėtų atlikti elektromagnetinės
vietos tyrimą. Jei išmatuotas lauko stipris toje vietoje, kurioje naudojama kietinimo lempa, viršija taikomą aukščiau nurodytą RF atitikties lygį, belaidė VALO kietinimo
lempa turi būti stebima, kad patvirtintumėte normalų veikimą. Jei pastebimas nenormalus veikimas, gali prireikti papildomų priemonių, pavyzdžiui, iš naujo orientuoti
arba perkelti kietinimo lempą.
bDažnių diapazone nuo 150 kHz iki 80 MHz lauko stiprumas turi būti mažesnis nei 3 V/m
a

190

191

VALO™ GRANDMTDAWL TAT- TQADDID

Deskrizzjoni tal-prodott• Biex tevita li tagħmel ħsara lit-tagħmir, M'GĦANDEKX iddaħħal is-swaba, strumenti, jew oġġetti oħra fil-kompartiment tal-batterija tad-dawl tat-tfejjaq.
• Biex tevita li tagħmel ħsara lit-tagħmir, M'GĦANDEKX tipprova tnaddaf il-kuntatti tad-deheb, jew kwalunkwe parti tal-kompartiment tal-batterija. Ċempel is-Servizz tal-Klijent
Ultradent jekk ikun hemm tħassib.
• Biex jiġi evitat ir-riskju ta' kontaminazzjoni inkroċjata, il-kmiem tal-barriera huma użu minn pazjent wieħed.
• Biex tnaqqas ir-riskju ta 'korrużjoni, neħħi l-kmiem tal-barriera wara l-użu.
• Biex jitnaqqas ir-riskju ta' reżini li jiċċekkjaw inqas, tużax dawl għat-tfejjaq jekk il-lenti tkun bil-ħsara.

Bl-ispettru tal-broadband tiegħu, id-dawl tat-tfejjaq tal-VALO Grand huwa ddisinjat biex polimerizza l-prodotti kollha li jkuraw id-dawl fil-firxa tat-tul ta 'mewġ ta' 385-515nm għal kull ISO
10650. Id-dawl tat-tfejjaq VALO Grand juża l-batteriji rikarikabbli Ultradent VALO u l-ċarġer tal-batterija. Id-dawl tat-tfejjaq huwa ddisinjat biex jistrieħ f'parentesi standard ta 'unità dentali
jew jista' jiġi mmuntat apposta bl-użu tal-parentesi inkluża mal-kit.

VALO Dawl ta’ Lqigħ Mingħajr Fili:

VALO Grand Product Components:

Uża

•
•
•
•
•
•

1) Kull modalità ta’ enerġija tintuża għat-tqaddid ta' materjali dentali b'inizjattivi tar-ritratti. Ara l-Gwida għall-Modalità Quick għall-ħinijiet irrakkomandati tat-tqaddid.
NOTA: Id-dawl tat-tqaddid huwa pprogrammat biex jiċċikla mill-Qawwa Standard għall-High-Qawwa Plus sal-modalità Qawwa Ekstra f'sekwenza. Pereżempju, biex tinbidel mill-modalitàStandard ta’ Qawwa għall-modalità Qawwa Ekstra, huwa meħtieġ li ċiklu fil-modalità ta' Qawwa Għolja Plus u mbagħad għall-modalità Qawwa Ekstra.
2) Id-dawl ta’ tqaddid jaħżen l-intervall tal-ħin u l-modalità l-aktar użati riċentement, u jonqos lura għal dan kull meta l-modalitajiet jinbidlu jew jekk il-batteriji jitneħħew.

1 — Dawl għat-tfejjaq ta' VALO Grand4 - Batteriji rikarikabbli ULTRADENT VALO
1 - Ċarġer tal-batterija ULTRADENT VALO b'adapter tal-enerġija 12VDC AC tal-grad mediku
1 - Pakkett tal-kampjun tal-Barrier Sleeve VALO
1 — VALO Tarka tad-dawl kbir
1 - Tfejjaq parentesi ta 'immuntar tal-wiċċ ħafif b'tejp adeżiv tal-stick doppju

Overview of Controls:

Ritratt

Operazzjoni
MOD TA’ KURA Modalità Enerġija Standard
Buttuni u Dwal

Deskrizzjoni

Buttuna / i tal-Qawwa(Fuq jew isfel)
Timing Dwal
Batterija Baxxa - Indikatur
Dawl tal-Modalità/Status
Buttuna li Tibdel il-Ħin /Modalità(Il-qiegħ)(Fuq)

Għall-prodotti kollha deskritti, aqra sew u tifhem l-istruzzjonijiet kollha u l-informazzjoni SDS qabel l-użu.

Indikazzjonijiet għall-Użu / Għan Maħsub

Is-sors ta’ illuminazzjoni għall-ikkurar ta' materjali u sustanzi adeżivi ta’ restorazzjoni dentali foto-attivati.

Twissijiet u Prekawzjonijiet
Grupp ta’ Riskju 2
ATTENZJONI UV emessa minn dan il-prodott. Irritazzjoni tal-għajnejn jew tal-ġilda tista’ tirriżulta minn espożizzjoni. Uża lqugħ xieraq.
ATTENZJONI Radjazzjoni ottika possibilment perikoluża emessa minn dan il-prodott. Tiħux fil-lampa tat-tħaddim. Jista’ jkun ta' ħsara għall-għajnejn.
• M'GĦANDEKX tħares direttament lejn il-ħruġ tad-dawl. Il-pazjent, il-kliniku, u l-assistenti għandhom dejjem jilbsu protezzjoni tal-għajnejn UV kkulurita ambra meta d-dawl tat-tfejjaq
ikun qed jintuża.
• Biex jiġi evitat ir-riskju ta 'xokk elettriku, l-ebda modifika ta' dan it-tagħmir mhi permessa. Uża biss il-provvista tal-enerġija Ultradent VALO inklużi u l-adapters tal-plagg. Jekk dawn
il-komponenti huma bil-ħsara, tużax u ċempel lil Servizz tal-Klijent Ultradent biex tordna sostituzzjoni.
• Tagħmir tal-komunikazzjonijiet RF portabbli jista 'jiddegrada l-prestazzjoni jekk jintuża eqreb minn 30 ċm (12 in.).
• Uża biss aċċessorji awtorizzati, kejbils, u provvisti ta 'enerġija biex jipprevjenu tħaddim mhux xieraq, żieda fl-emissjonijiet elettromanjetiċi jew tnaqqis fl-immunità elettromanjetika(irreferi għat-taqsima tal• Biex jiġi evitat ir-riskju ta' nar elettriku assoċjat mal-immaniġġjar tal-batteriji:⸰ Tħallatx batteriji rikarikabbli ma' batteriji mhux rikarikabbli jew tipi oħra ta' batteriji.
⸰ M'GĦANDEKX tipprova tiċċarġja batteriji mhux rikarikabbli. L-iċċarġjar ta 'batteriji mhux rikarikabbli jista' jirriżulta f'korriment jew ħsara fil⸰ M'għandekx awtoklav jew isprej batteriji, kuntatti tal-batterija, ċarġer, jew adapter tal-enerġija AC b'likwidu ta 'kwalunkwe tip. Jekk tidher il-korrużjoni fuq il-kuntatti tal-ċarġer
tal-batterija, ċempel Servizz tal-Klijent Ultradent biex tordna sostituzzjoni.
⸰ M'GĦANDEKX iċċarġja batteriji madwar materjali li jaqbdu.
⸰ M'GĦANDEKX iżomm iċ-ċarġer f'operatorju kliniku.
• Biex tevita r-riskju ta 'korriment, TUŻAX batteriji li huma mkorrudija (sadid), dented, jarmu riħa jew fluwidi, għandhom tgeżwir imqatta' jew nieqes, jew inkella huma mħassra.
Ċempel Servizz tal-Klijent Ultradent biex tordna batteriji ta 'sostituzzjoni• Uża biss batteriji rakkomandati. Batteriji alternattivi jistgħu jikkawżaw malfunzjonament.
• Biex tipprevjeni r-riskju ta' irritazzjoni termali jew korriment, evita ċikli ta' tfejjaq lura għal lura u tesponix tessuti rotob orali f'qrib għal aktar minn 10 sekondi fi kwalunkwe modalità.
Jekk huma meħtieġa ħinijiet itwal tat-tfejjaq, uża ċikli multipli tat-tfejjaq b'perjodi ta' mistrieħ bejn iċ-ċikli, jew uża prodott b'kura doppja biex tevita li jsaħħan it-tessut artab.
• Oqgħod attent meta tikkura pazjenti li jbatu minn reazzjonijiet jew sensittivitajiet fotobijoloġiċi avversi, pazjenti li jkunu għaddejjin minn kura bil-kimoterapija, jew pazjenti li jkunu
qed jiġu kkurati b'medikazzjoni fotosensitizzanti.
• Din l-unità tista' tkun suxxettibbli għal kampi elettriċi manjetiċi jew statiċi qawwija, li jistgħu jfixklu l-ipprogrammar. Jekk tissuspetta li dan seħħ, aqbad l-unità momentarjament umbagħad erġa' plaggjaha fl-iżbokk.
• M'GĦANDEKX imsaħ id-dawl tat-tqaddid b'tindif kawstiċi jew joborxu, awtoklavi, jew għaddas fi kwalunkwe tip ta 'banju ultrasoniku, diżinfettant, soluzzjoni tat-tindif, jew likwidu.
Nuqqas ta' sekwenza tal-istruzzjonijiet tal-ipproċessar, inklużi, jistgħu jirrendu d-dawl tat-tqaddid mhux operabbli.
192

Il-VALO Mingħajr Fili Light Shield huwa f'forma ovali, jista’ jiddawwar għall-użu massimu, u jista' jintuża bi kmiem trasparenti tal-barriera..

L-INTERVALI TAŻ-ŻMIEN: 5, 10, 15, u20 sekonda.
• Id-dawl tat-tqaddid jonqos għal din il-modalità meta JINĦALLA INIZJALMENT. Id-Dawl tal-Modalità / Status se jkun aħdar u l-erba 'Dwal ta' Timing ħodor huma mdawla, li jindikaw
il-modalità Standard ta’ Enerġija.
• Biex tbiddel l-intervalli tal-ħin, agħfas malajr il-Buttuna Ħin / Modalità.
• Agħfas il-buttuna ta’ l-enerġija biex tfejjaq. Biex twaqqaf it-tqaddid qabel ma jitlesta l-intervall tal-ħin, erġa’ agħfas il-Buttuna tal-Qawwa.

MOD TA’ KURA Modalità ta’ Qawwa Għolja Plus

L-INTERVALI TAŻ-ŻMIEN: 1, 2, 3, u 4 sekondi.
• Mill-modalità Qawwa standard, agħfas u żomm il-Buttuna tal-Bidla tal-Ħin / Mod għal 2 sekondi u erħi. Id-Dawl tal-Modalità / Status se jkun oranġjo, u l-erba 'Dwal Ħodor ta' Timing
idawlu u jixgħelu, li jindikaw modalità ta’ Qawwa Għolja.
• Biex tbiddel l-intervalli tal-ħin, agħfas malajr il-Buttuna Ħin / Modalità.
• Agħfas jew il-Buttuna tal-Qawwa biex tfejjaq. Biex twaqqaf it-tqaddid qabel ma jitlesta l-intervall tal-ħin, erġa’ agħfas jew il-Qawwa Button.
• Biex terġa’ lura għall-modalità Qawwa standard, agħfas u żomm il-Buttuna tal-Bidla fil-Ħin / il-Mod għal 2 sekondi u erħi, dan iċ-ċiklu għall-modalità Qawwa Ekstra. Agħfas u żommgħal darb'oħra għal 2 sekondi, u erħi. Id-Dawl tal-Modalità / Status se jkun aħdar u l-erba 'Dwal Ħodor ta' Timing huma mdawla, li jindikaw il-Modalità Standard.

MOD TA’ KURA Modalità Qawwa Ekstra

INTERVAL TAŻ-ŻMIEN: 3 sekondi biss (Nota: Il-modalità Qawwa Ekstra għandha dewmien ta’ sigurtà ta' 2 sekondi fl-aħħar ta’ kull ċiklu ta' tqaddid biex tillimita t-tisħin waqt tqaddid
konsekuttiv. Fl-aħħar tad-dewmien, il-ħoss jindika li l-unità hija lesta għal użu kontinwu).
• Mill-Modalità Standard ta’ Qawwa, agħfas il-Buttuna tal-Bidla fil-Ħin / Mod għal 2 sekondi, erħi, agħfas u żomm mill-ġdid għal 2 sekondi, u erħi. Id-Dawl tal-Modalità / Status se jkunoranġjo u jteptep, u tlieta mill-Ħodor Timing Lights ser idawwal u jixgħelu, li jindikaw il-mod Qawwa Ekstra.
• Agħfas il-buttuna ta’ l-enerġija biex tfejjaq. Biex twaqqaf it-tqaddid qabel ma jitlesta l-intervall tal-ħin, erġa’ agħfas il-Buttuna tal-Qawwa.
• Biex terġa’ lura għall-modalità Qawwa standard, agħfas u żomm il-Buttuna Ħin / Mod għal 2 sekondi u erħi. Id-Dawl tal-Modalità / Status se jkun aħdar u l-Ħodor Id-Dwal tat-Timing
huma mdawla, li jindikaw il-Modalità Standard ta’ Qawwa.

Modalità ta’ Rqad: Id-dawl tat-tqaddid jidħol fil-Modalità Irqadwara 60 sekonda ta’ inattività, kif indikat minn li jteptep bil-mod tad-dawl tal-modalità / status. Meta taqbad jew tmiss ma'l-unità, tqum id-dawl ta' tqaddid u terġa’ lura awtomatikament għall-aħħar setting użat. Biex timmassimizza l-ħajja tal-batterija, ħalli d-dawl tal-ikkurar mhux disturbat meta ma jkunx
qed jintuża.

Naddaf

1) Armi l-kmiem tal-barriera użati fi skart standard wara kull pazjent.
2) Ara t-Taqsima tal-Ipproċessar.

Struzzjonijiet tal-Brekit tal-Immuntar

1) Il-brekits għandhom jiġu mmuntati fuq wiċċ ċatt u bla żejt.
2) Wiċċ nadif bl-alkoħol tat-tħakkik.
3) Qaxxar irfid it-tejp li jwaħħal tal-brekit.
4) Poğği l-brekit sabiex id-dawl ta’ tqaddid jogħla' l fuq meta jitneħħa. Agħfas sew f'postha.

193

Gwida għall-Modalità Quick:Qawwa standard
1000 mW/cm2

ModalitàHigh Qawwa Plus
1600 mW/cm2

Qawwa Ekstra
3200 mW/cm2

Buttuna tal-Enerġija

Kif tiċċarġja l-batteriji:
1) Poġġi ċarġer u batteriji f'żona fejn dawn mhux se jiġu f'kuntatt ma 'likwidi jew sprej diżinfettant.
2) Daħħal batteriji fil-ċarġer b'tarf pożittiv (+) ippontat lejn id-dwal indikatur fuq iċ-ċarġer.
3) Id-Dwal Aħdar jindikaw li l-batteriji huma lesti għall-użu.
4) Il-batteriji jieħdu 1-3 sigħat biex jiċċarġjaw. Ħalli l-batteriji fil-ċarġer sakemm ikunu lesti għall-użu.
NOTA: Jekk id-dawl aħmar fuq il-ċarġer ma jdurx aħdar meta l-batteriji jkunu ilhom jiċċarġjaw itwal minn tliet sigħat, il-batterija tista 'tkun degradata (korrotta) u ma tistax tiġi ċċarġjata.
Ipprova batterija ġdida jew ċempel Servizz tal-Klijent Ultradent biex tordna sett ġdid ta 'batteriji rikarikabbli.

LEDs tal-Modalità / Timing

Buttuni ĦinĦin Għażliet

5s

10s

15s

20s

1s

2s

3s

4s

3s Biss

Biex Tibdil ĦinAgħfas u erħi l-Buttuna tal-Ħin malajr biex iddur l-għażliet tal-ħin.

Biex Ibdel ModiAgħfas u żomm il-Buttuna tal-Ħin għal 2 sekondi u ħoll. Id-dawl ta’ tqaddid jaħdem għal Modalità li jmiss.

Deskrizzjoni

LEDs solidi

Teptip LEDs

Gwida ta’ Tqaddid Quick:
Modalità tat-Tfejqan

Qaw wa * (mW)

Irradjanza * (mW/c m2)

Ħin Totali ta 'Espożizzjoni (Sekondi)Enerġija** (Joules)

Standard

970

900

5/10/15/20

4.8J/9.7J/14.5J/19.4J

Qawwa Għolja Plus

1,620

1,500

1/2/3/4

1.6J/3.2J/4.8J/6.5J

Xtra

2,260

2,100

3

6.8J

Gwida ta’ Twissija ta’ Qasira:2) Miters tad-dawl huma differenti ħafna u huma ddisinjati għal ponot speċifiċi ta' gwida tad-dawl u lentijiet. Ultradent jirrakkomanda li jiċċekkja rutina l-output fil-modalità Standard
Power. NOTA: il-ħruġ numeriku veru se jkun imkeb minħabba l-eżattezza ta 'meters tad-dawl komuni u l-pakkett LED tad-dwana fid-dawl tat-tfejjaq
Manutenzjoni tal-BatterijaIċċarġjar u Bidla tal-Batteri
Id-dawl tat-tfejjaq jiġi b'4 batteriji tal-fosfat tal-ħadid tal-litju rikarikabbli. Ultradent jirrakkomanda li tuża batteriji rikarikabbli VALO biss.

* Ħruġ nominali f'distanza ta' 2 mm mill-ponta tal-lenti sal-wiċċ ta' fuq tal-kompost.
**Il-valuri fit-taqsima tal-enerġija totali għal kull ċiklu (joules) huma arrotondati sal-eqreb għaxra.

Twissija tal-Livell ta’ Qawwa

Twissija tat-Temperatura

Twissija tas-Servizz

Twissija LED

Ibdel il-batterijiĦalli jiksaħ

Ċempel lill-Customer Service għalTiswijaĊempel lill-Customer Service għalTiswija• L-ebda ħoss• Teptip kull 2 sekondi• Tippermetti l-operazzjoni• 3 beeps kontinwi• Flash wieħed rapidu• Tipprojbixxi operazzjonijiet• Batterija baxxa: iteptep bil-mod• Itfi: 3 beeps, li jteptep• Tipprojbixxi l-operazzjoni• 3 beeps• Teptip bil-mod• Tipprojbixxi l-operazzjoni

Kif tibdil/daħħal il-batteriji:
1) Neħħi l-għatu tad-dahar billi tgħawweġ kontra l-arloġġ kwart dawra.
2) Neħħi l-batteriji.
3) Daħħal l-ewwel naħa pożittiva (+) tal-batteriji friski.
4) Erġa' waħħal l-għatu ta' wara billi tallinja u timbotta bil-mod waqt li tgħawweġ lejn il-lemin. L-għatu tikklikkja meta mwaħħal bis-sħiħ.
5) L-unità hija lesta għall-użu.
Batteriji Baxxi: Id-dawl tat-tfejjaq jissinja lill-utent li wasal iż-żmien li tbiddel il-batteriji meta d-Dawl tal-Indikatur tal-Batterija Baxxa qed iteptep aħmar. Jekk il-ħlas tal-batterija jsir baxx
wisq, se jseħħ ħoss ta 'twissija ta' biep 3 li jinstema' u d-dawl tat-tfejjaq ma jippermettix aktar tħaddim sakemm il-batteriji jiġu ċċarġjati mill-ġdid jew jiddaħħlu batteriji ġodda. (Aral-Gwida ta 'Twissija Malajr)
VALO bla fil/VALO Grand BISS (UD BISS)
Ħin tal-Iċċarġjar u l-Ħajja tal-Batterija: Il-ħajja tal-batterija ċċarġjata kompletament fid-dawl tat-tfejjaq hija dipendenti fuq l-intervall tal-Mod/Ħin, it-tip tal-batterija, l-ammont ta 'użu, ul-effiċjenza LED.
• Intervall Recharge rakkomandat: Recharge kif meħtieġ jew, meta indikatur batterija baxxa jidħol fuq. L-istennija tal-ħajja tiddependi fuq l-użu.
• Batteriji Extra: Id-dawl tat-tfejjaq jiġi b'4 batteriji rikarikabbli.
Tiswija tal-Manifattur
1) It-tiswijiet għandhom isiru biss minn persunal tas-servizz awtorizzat. Ultradent huwa li jipprovdi lill-persunal tas-servizz b'dokumentazzjoni biex iwettaq tiswijiet.
2) Meta tibgħat unitajiet fil għat-tiswija, servizz, jew kalibrazzjonijiet dejjem neħħi l-batteriji mid-dawl tat-tfejjaq u l-ċarġer. Wrap batteriji, ċarġer, adapter, u d-dawl tat-tfejjaq
separatament fil-kaxxa tar-ritorn.
3) Batteriji tal-vapur skont ir-regolamenti lokali.
Garanzija
Ultradent Products, Inc. (“Ultradent”) jiggarantixxi li dan il-prodott għandu, għal perjodu ta' 5 snin mid-data tax-xiri, meta jitħaddem skont l-istruzzjonijiet operattivi inklużi mal-prodott,(i) jikkonforma fl-aspetti materjali kollha mal-ispeċifikazzjonijiet stabbiliti fid-dokumentazzjoni ta' Ultradent li takkumpanja l-prodott; u (ii) ikun ħieles minn difetti fil-materjal u l-abbiltà.
Din il-garanzija limitata mhix trasferibbli u tapplika biss għax-xerrej oriġinali u ma testendix għas-sidien sussegwenti tal-prodott. Din il-garanzija limitata ma tkoprix l-ebda komponentiaċċessorji oħra bħal, iżda mhux limitata għal, batteriji, ċarġers, adapters, jew lentijiet adattivi. Din il-garanzija limitata hija nulla jekk il-prodott jonqos jew ikun bil-ħsara minħabba
negliġenza, abbuż, użu ħażin, inċident, modifika, tbagħbis, alterazzjoni, jew nuqqas li jsegwu l-istruzzjonijiet applikabbli għall-użu. Għal skopijiet ta' eżempju biss, prodott li jitwaqqa' u
jkun bil-ħsara mhuwiex kopert taħt din il-garanzija. Biex tikkwalifika taħt din il-garanzija limitata, prova tax-xiri (eż., irċevuta tal-bejgħ jew dokumentazzjoni simili) trid tiġi sottomessa lil
Ultradent flimkien mal-prodott difettuż.
Prodott difettuż li jissodisfa l-kundizzjonijiet tal-garanzija stabbiliti hawnhekk se, fid-diskrezzjoni unika ta 'Ultradent, jew jissewwa jew jinbidel. Fl-ebda każ m'għandha r-responsabbiltà
ta' Ultradent għall-prodott taqbeż il-prezz tax-xiri mħallas mix-xerrej. Taħt l-ebda ċirkostanza Ultradent ma għandu jkun responsabbli għal kwalunkwe danni indirett, inċidentali, previst,
mhux previst, speċjali, jew konsegwenzjali li jirriżultaw minn jew b'rabta mal-użu ta' dan il-prodott.

Ipproċessar

Dan il-prodott huwa maħsub biex jintuża b'kmiem ta' barriera. Ara hawn fuq għall-manutenzjoni u t-tindif ġenerali.

Ħażna u Rimi

Jekk taħżen id-dawl tat-tqaddid għal perjodi itwal minn ġimgħatejn, jew tippakkjaha għall-ivvjaġġar, dejjem neħħi l-batteriji. Jekk il-batteriji jitħallew fl-unità għal perjodi twal ta’ żmien
mingħajr iċċarġjar jistgħu jsiru mhux funzjonali jew ma jistgħux jiġu ċċarġjati. Taħżinx il-batteriji f'temperaturi 'l fuq minn 40°C (104°F) jew f'dawl dirett tax-xemx.

Manutenzjoni

Tindif Ġenerali tad-Dawl tat-Tfejjaq
Wara kull użu, niedja garża jew drapp artab b'diżinfettant tal-wiċċ approvat u imsah il-wiċċ u l-lenti. It-tindif mhux awtorizzat jista 'jikkawża ħsara lid-dawl tat-tqaddid.
TINDIF AĊĊETTABBLI:• 70% alkoħol isopropil• 70% Etanol

Tindif Ġenerali tat-Tarka tad-Dawl:
Naddaf it-tarka tad-dawl VALO mingħajr fili billi tuża kwalunkwe diżinfettant tal-wiċċ. M'GĦANDEKX awtoklava.
sewwi
Manutenzjoni Mwettqa mill1) Uża kmiem tal-barriera biex tipprevjeni li l-kompost dentali jeħel mal-wiċċ tal-lenti. Jekk meħtieġ, uża strument dentali tal-plastik jew tal-istainless steel biex tneħħi bir-reqqa
kwalunkwe kompost adeżerit. Tużax għodod li jagħmlu ħsara lill-lenti.
194Ħadid tqaddid Ħażna u Trasport:• Temperatura: + 10 ° C sa + 40 ° C (+ 50 ° F sa + 104 ° F)
• Umdita relattiva: 10% sa 95%• Pressjoni tal-Ambjent: 500 hPa sa 1060 hPa
Meta tarmi skart elettroniku (jiġifieri dwal ta’ tqaddid, ċarġers, batteriji u provvisti ta’ enerġija), segwi l-linji ta’ gwida lokali dwar l-iskart u r-riċiklaġġ.

Konsiderazzjonijiet Tekniċi
Aċċessorji
Punt

Informazzjoni CE

VALO Lqugħ tad-dawl mingħajr fili

195

Punt

Informazzjoni CE

Informazzjoni teknika / Data

Meded ta’ Wavelength li Jikkuraw Effettivi Komposti:

VALO ĊarġerMEDED EFFETTIVI TAT-TUL TAL-MEWĠ TAT-TKURAĠĠ
(Firxa ta 'Wavelength Użabbli: 385-515 nm)
QAWWA RADJANTI SPETTRALI (mW/nm)

Batteriji VALO

Effettiv aktar baxx
medda tal-mewġ
395-415 nm

Faxxa effettiva
tal-mewġ ta
440-480nm

Xtra
Qawwa Għolja
Standard

WAVEL LENG TH (nm)Attribut

Informazzjoni / Speċifikazzjoni

Lenti

Dijametru 11.7 mm

Medda ta’wavelength• Medda utilizzabbli ta’ wavelength: 385 - 515nm• Tul ta’ mewġiet l-aktar għoljin: 395 - 415nm u 440 - 480nm
Tabella ta’ Tqabbil ta' Ħruġ Radjali Nominali
Strument tal-Kejl

Tabella ta’ Intensità
tad-Dawl

196‡ Analy Analizzatur tal-ispettru ta’ Gigahertz
Ħruġ

Qawwa totali

Apertura tal-Meter

15 mm

15 mm

Qawwa standard(±10%)

900 mW/cm²

970 mW

High Qawwa Plus(±10%)

1500 mW/cm²

1615 mW

Qawwa Ekstra(±10%)

2100 mW/cm²

Radiant Exitance se tvarja abbażi tal-kapaċità
tal-istrument, il-metodu tal-kejl, u t-tqegħid
tad-dawl.
‡ Eżitanza Radjanti tikkonforma mal-ISO
10650 meta titkejjel b'analizzatur tal-ispettru
Gigahertz-Optik.

2260 mW

VALO Dawl Dukat
tat-Tqaddid

Klassifikazzjonijiet: IEC 60601-1 (Sikurezza), IEC
60601-1-2 (EMC)Piż:• Bil-batteriji: 6 oz. (190 gramma)• Mingħajr batteriji: 5 oz. (150 gramma)
Dimensjoni: (8 x 1.28 x 1.06) pulzieri, (203 x 32.5 x 27) mm

Ċarġer Provvista ta’
l-Elettriku

Output - 12VDC f’500mA
Input - 100VAC sa 240VAC
Ultradent P / N 5930 VALO Ċarġer Provvista ta’
l-Elettriku b' plaggs internazzjonali

Klassifikazzjonijiet: IEC 60601-1 (Sigurtà)
Tul tal-kurdun - 6 piedi (1.8 metri)
VALO Ċarġer Qawwa Supply huwa provvista ta’ enerġija tal-Klassi Medika
tal-Klassi II Mediku u jipprovdi iżolazzjoni mill-MAINS Qawwa

197

Informazzjoni mixxellanja

Attribut

Informazzjoni / Speċifikazzjoni

Dikjarazzjoni ta’ Gwida u Manifattura għal Emissjonijiet Elettromanjetiċi

VALO Ċarġer

Ċarġer tal-batterija intelliġenti tal-litju tal-fosfat tal-litju VALO 3.6VDC:• Tifi awtomatiku meta ċċarġjat kompletament• Sejbien awtomatiku ta’ batteriji difettużi
• Protezzjonijiet: Termali, Overcharge, Short-circuit, polarità b'lura⸰ LED aħmar - Iċċarġjar
⸰ LED aħdar - vojt jew iċċarġjat għal kollox
⸰ LED mixgħul - short circuit• Ħin tal-iċċarġjar: 1 - 3 sigħat
Рэйтынг: CE, WEEE

Emissjonijiet Test

Konformità

Ambjent elettromanjetiku - gwida

Emissjonijiet RF
CISPR 11

Grupp 1

Id-dawl ta’ tqaddid juża batteriji u mhux affettwat minn EMI, RF, jew trażżin ta' mewġ.

Emissjoni RF
CISPR 11

Klassi B

Id-dawl tat-tqaddid juża enerġija elettrika u elettromanjetika għall-funzjonijiet interni tagħhom
biss. Għalhekk, kwalunkwe emissjonijiet RF huma baxxi ħafna u mhux probabbli li jikkawżaw
interferenza fit-tagħmir elettroniku fil-qrib.

Batteriji VALO

Rikkarikabbli: Kimika sikura Fosfat tal-Ħadid tal-Litju (LiFePO4) RCR123A•
Vultaġġ tax-Xogħol: 3.2VDC, 400mAh (Watt siegħa meħtieġa)Klassifikazzjonijiet: UL 62133-2, CE, RoHS, WEEE
Iċ-ċelloli huma kklassifikati għa> 2000 ċiklu ta 'ċarġja/kwittanza qabel ma jolqtu 70% tal-kapaċità tiegħu. Huwa rakkomandabbli lil-batteriji jiġu sostitwiti kull 3-5 snin, iżda l-ħajja tista 'tvarja skont l-użu. Ibdel kif meħtieġ.

Emissjonijiet armoniċi
IEC 61000-3-2

M/A

L-ittestjar tal-emissjonijiet armoniċi u tal-fluttwazzjonijiet tal-vultaġġ mhumiex applikabbli għaddawl tal-ikkurar minħabba li jaħdmu bil-batterija.

Kondizzjonijiettal-Operat

Temperatura: + 10 ° C sa + 32 ° C (+ 50 ° F sa + 90 ° F)
Umdita relattiva: 10% sa 95%
Pressjoni tal-Ambjent: 700 hPa sa 1060 hPa

Varjazzjonijiet tal-vultaġġ /emissjonijiet ta’ ċaqliq IEC
61000-3-3

M/A

Id-dawl ta’ tqaddid huwa adattat għall-użu fl-istabbilimenti kollha, inklużi stabbilimenti domestiċi
u dawk konnessi direttament man-netwerk pubbliku ta' provvista ta’ enerġija ta' vultaġġ baxx li
jipprovdi bini għal użu domestiku.

Ċiklu tax-Xogħol

Id-dawl tat-tqaddid huwa ddisinjat għal tħaddim għal żmien qasir. F'temperatura massima ta’ l-ambjent (32 ° C) minuta waraċikliżmu ta' wara ma 'dahar, 30 minuta OFF (perjodu ta' tkessiħ).
Dikjarazzjoni ta’ Gwida u Manifattura għall-Immunità Elettromanjetika

Touble Shooting
Jekk is-soluzzjonijiet issuġġeriti hawn taħt ma jirranġawx il-problema, jekk jogħġbok ċempel lil Ultradent f '800.552.5512. Barra l-Istati Uniti, ċempel lid-distributur
Ultradent tiegħek jew lin-negozjant tas-snien.
Problem

198

Id-dawl tat-tqaddid huwa maħsub għall-użu fl-ambjent elettromanjetiku speċifikat hawn taħt. Il-klijent jew l-utent għandu jiżgura li jintuża f'ambjent bħal dan.

Possible Solutions

Id-dawl ma jinxtegħelx

1) Agħfas il-Ħin / Modalità jew il-Buttuna ta’ l-Enerġija biex tqum mill-Modalità Qawwa Save.
2) Iċċekkja l-Indikatur tal-Batterija Baxx L-aħmar għall-istatus tal-iċċarġjar tal-batterija.
3) Iċċekkja li batteriji friski huma mdaħħla sewwa fl-unità.
4) Jekk LEDs ta’ Twissija ħomor u sofor jteptpu dan ifisser li d-dawl ta' tqaddim laħaq il-limitu ta’ sigurtà fit-temperaturainterna tiegħu. Ħalli d-dawl ta’ tqaddid jiksaħ għal 10 minuti jew uża xugaman niedi niexef biex jiksaħ l-unità malajr.
5) Jekk LED aħmar tat-Twissija jixgħel u ħoss kontinwament, ċempel lil Ultradent Customer Service għal tiswija.

Id-dawl ma jibqax mixgħul għall-ħin
mixtieq1) Iċċekkja l-Modalità tad-Dwal u d-Dwal għall-input korrett tal-ħin.
2) Iċċekkja l-Indikatur tal-Batterija Baxx għall-istatus tal-iċċarġjar tal-batterija.
3) Iċċekkja li batteriji friski huma mdaħħla sewwa fl-unità.

Dawl mhux tqaddid reżini sew1) Iċċekkja l-lenti għal raża / kompożiti kkurati residwi.
2) Bl-użu ta’ protezzjoni ambra UV ta' l-għajnejn sewwa, ivverifika li d-dwal LED qed jaħdmu.
3) Iċċekkja l-livell tal-enerġija bil-miter tad-dawl. Jekk tuża miter tad-dawl, Ultradent jirrakkomanda li tiċċekkja d-dawl ta’
tqaddid fil-modalità Qawwa standard.
NOTA: L-output numeriku veru se jkun mgħawweġ minħabba l-ineżattezza tal-miters tad-dawl komuni u l-LED tad-dwana
jippakkja l-użi tad-dawl tat-tqaddid. Miters tad-dawl ivarjaw ħafna u huma ddisinjati għal ponot u lentijiet speċifiċi ta’ gwida
tad-dawl.
4) Iċċekkja d-data ta’ skadenza fuq ir-raża ta’ tqaddid
5) Kun żgur li qed tiġi segwita teknika xierqa skont ir-rakkomandazzjonijiet tal-manifattur.

Batteriji mhux se jiċċarġjaw

1) Kun żgur li l-batteriji jiddaħħlu fiċ-ċarġer fl-orjentazzjoni korretta u ħalli l-batteriji jiċċarġjaw għal 1-3 sigħat.
2) Jekk id-dwal ħomor fuq il-ċarġer ma jinbidlux għal aħdar, ċempel lis-Servizz tal-Klijent Ultradent biex tordna batteriji u /jew ċarġer sostituti.
3) Jekk la dwal ħodor u lanqas ħomor fuq iċ-ċarġer ma huma viżibbli, ċempel lis-Servizz tal-Klijent Ultradent biex tordna jew
ibdel iċ-ċarġer u / jew l-adapter AC.

Iċ-ċarġer ma jiċċarġjax il-batteriji

1) Kun żgur li ċ-ċarġer huwa pplaggjat u l-adapter AC huwa pplaggjat fi plakka tal-enerġija li taħdem.
2) Jekk id-dwal ħodor jew ħomor fuq iċ-ċarġer mhumiex viżibbli, ċempel lil Ultradent Customer Service għal ċarġer ġdid u/ jew adapter AC.

Ma tistax tbiddel il-mod jew l-intervalli
tal-ħin1) Żomm kemm il-buttuni tal-Ħin / Modalità kif ukoll ta’ l-Enerġija' l isfel sakemm serje ta’ ħsejjes jindikaw li d-dawl tattqaddid ma jinfetaħx.

Id-dawl tat-tqaddid huwa maħsub għall-użu fl-ambjent elettromanjetiku speċifikat hawn taħt. Il-klijent jew l-utent għandu jiżgura li jintuża f'ambjent bħal dan.
Test tal-IMMUNITÀ

IEC 60601 livell tat-test

Livell ta’ konformità

Gwida dwar l-ambjent elettromanjetiku

Kwittanza elettrostatika (ESD)

Kuntatt ± 8 kV
Arja ± 15 kV

Kuntatt ± 8 kV
Arja ± 15 kV

L-ambjent fiżiku għandu jkun ristrett għal dan li ġej:1) Kodiċi IP: IP20
2) Tgħaddisx fil-likwidu.
3) Tużax madwar gass li jaqbad. Unità mhix APG
u mhux AP.
4) Medda ta’ umdità tal-ħażna: 10% - 95%
5) Il-firxa tat-temperaturi tal-ħażna: 10 ° C - 40 ° C

Elettriku mgħaġġel temporanju/ fqigħ
IEC 61000-4-4± 2 kV għal-linji tal-provvista
tal-enerġija
± 1 kV għal-linji ta’ dħul / ħruġ± 2 kV għal-linji tal-provvista
tal-enerġija
Nota 1: Id-dawl tat-tqaddidm'għandux portijiet I / 0

Id-dawl tat-tqaddid jaħdem bil-batterija u ma jkunxkapaċi li jgħaqqad l-AC MAINS Qawwa.

Iż-żieda
IEC 61000-4-5± 1 kV linja għall-linja
Linja ta’ ± 2 kV mad-dinja± 1 kV linja għall-linja
Linja ta’ ± 2 kV mad-dinja

Vultaġġ, rjus, xorts,
interruzzjonijiet u varjazzjonijiet
fil-linji tad-dħul tal-provvista ta’
l-enerġija

Iċ-ċarġer tal-batterija aċċessorju jista’ jkun suġġett
<5% U
<5% U
(> 95% għaddas f'U għal 0.5 ċiklu) (> 95% għaddas f'U għal 0.5 ċiklu) għal dan ta' hawn fuq iżda huwa separat minn u mhux
kritiku għall-operazzjoni tad-dawl ta’ tqaddid mingħajr
fili tal-VALO.
40% U
40% U
(60% dip f'U għal 5 ċikli)(60% dip f'U għal 5 ċikli)
Jekk il-vultaġġ tal-batterija tad-dawl ta’ tqaddid
jaqa' għal 4VDC l-unità ma tippermettix it-tħaddim.
70% U
70% U
Id-dawl tal-ikkurar tal-VALO mingħajr fili jintefa. Meta
(30% dip f'U għal 25 ċiklu)(30% dip f'U għal 25 ċiklu)
jiddaħħlu batteriji ġodda u jiġu restawrati l-livelli
xierqa ta’ l-enerġija, id-dawl ta' tqaddid jerġa’ jibda u
<5% U
<5% U
jerġa' lura għall-istess stat qabel it-telf ta’ l-enerġija.
(> 95% dip f'U għal 5 s)(> 95% dip f'U għal 5 s)
Id-dawl tat-tqaddid jirkupra waħdu fil-każ ta’ telf
Nota 2: Awto jirkupra
ta' enerġija.

IEC 61000-4-2

IEC 61000-4-11

Frekwenza tal-qawwa
Kamp manjetiku (50/60 Hz)

30 A/m30 A/mIEC 61000-4-8Minħabba li d-dawl ta’ tqaddid jaħdem bil-batterija,
mhuwiex suġġett għal transjenti elettriċi, surges,
vultaġġ baxx, xorts, interruzzjonijiet, jew varjazzjonijiet
fuq l-enerġija AC PRINĊIPALI.

PIl-kampijiet manjetiċi tal-frekwenza tal-enerġija
għandhom ikunu f'livelli karatteristiċi ta’ post tipiku
f'ambjent tipiku, residenzjali, tal-kura tas-saħħa fiddar, kummerċjali, tal-isptar, jew militari.

Nota 1: Id-dawl tat-tqaddid m'huwiex mgħammar b'xi portijiet jew linji I / O aċċessibbli.

199

Dikjarazzjoni ta’ Gwida u Manifattura għal Immunità ElettroManjetika għal sistemi ta’ appoġġ mhux tal-ħajja
Id-dawl tat-tqaddid huwa maħsub għall-użu fl-ambjent elettromanjetiku speċifikat hawn taħt. Il-klijent jew l-utent għandu jiżgura li jintuża f'ambjent bħal dan.
Test tal-IMMUNITÀ

IEC 60601 livell
tat-test

Konduzzjoni RF

3 Vrms3 VrmsIEC 61000-4-6

150 kHz sa 80 MHz

150 kHz sa 80 MHz

Livell ta’ konformità

Gwida dwar l-ambjent elettromanjetikuTagħmir ta’ komunikazzjoni RF portabbli u mobbli m'għandux jintuża mhux eqreb lejn
kwalunkwe parti tad-dawl ta' tqaddid, inklużi kejbils, mid-distanza ta’ separazzjoni
rakkomandata kkalkulata mill-ekwazzjoni applikabbli għall-frekwenza tat-trasmettitur.
Distanza ta’ separazzjoni rakkomandata:

80 MHz sa 800 MHz

800 MHz sa 2.5 GHz
RF irradjat

3V/m

3V/m

P hija l-ogħla rata tal-qawwa tal-ħruġ tat-trasmettitur f'watts (W) skont il-manifattur tattrasmettitur u d hija d-distanza ta’ separazzjoni rakkomandata f'metri (m).
Il-qawwiet tal-kamp minn trasmettituri RF fissi, kif iddeterminat minn stħarriġ dwar
is-sit elettromanjetiku, għandhom ikunu inqas mil-livell ta’ konformità f'kull rangeb
tal-frekwenza.

IEC 61000-4-3

80 MHz sa 2.5 GHz

80 MHz sa 2.5 GHz

Tista’ sseħħ interferenza fil-viċinanza tat-tagħmir immarkat bis-simbolu li ġej:NOTA 1 Fi 80 MHz u 800 MHz, tapplika l-iskala ta’ frekwenza ogħla.
NOTA 2 Dawn il-linji gwida jistgħu ma japplikawx fis-sitwazzjonijiet kollha. Il-propagazzjoni elettromanjetika hija affettwata mill-assorbiment u r-riflessjoni minnstrutturi, oġġetti u nies.
Qawwa tal-kamp minn trasmettituri fissi, bħal stazzjonijiet bażi għat-telefoni tar-radju (ċellulari / Mingħajr Fili) u radjijiet mobbli fuq l-art, radju dilettanti, xandir tarradju AM u FM u xandir televiżiv ma jistgħux jiġu mbassra teoretikament bi preċiżjoni. Sabiex jiġi stmat l-ambjent elettromanjetiku minħabba trasmettituri fissi tar-RF,għandu jiġi kkunsidrat stħarriġ fuq is-sit elettromanjetiku. Jekk il-qawwa tal-kamp imkejla fil-post li fih jintuża d-dawl ta’ tqaddid taqbeż il-livell ta' konformità RFapplikabbli hawn fuq, id-dawl ta’ tqaddid mingħajr fili tal-VALO għandu jiġi osservat biex jivverifika t-tħaddim normali. Jekk tkun osservata prestazzjoni mhux normali,
jistgħu jkunu meħtieġa miżuri addizzjonali, bħall-orjentazzjoni mill-ġdid jew ir-rilokazzjoni tad-dawl tat-tqaddid.
b
Fuq il-medda ta’ frekwenzi 150 kHz sa 80 MHz, il-qawwiet tal-kamp għandhom ikunu inqas minn 3 V / m.

200

Dikjarazzjoni ta’ Gwida u Manifattura għal distanzi ta' separazzjoni rakkomandati bejn tagħmir ta’ komunikazzjoni RF portabbli u mobbli u d-dawl ta' tqaddid mingħajr
fili VALO
Id-dawl tat-tqaddid huwa maħsub għall-użu f'ambjent elettro-manjetiku li fih jiġu kkontrollati disturbi RF irradjati. L-utent tad-dawl ta’ tqaddid jista' jgħin biex
jipprevjeni interferenza elettromanjetika billi jżomm distanza minima bejn tagħmir tal-komunikazzjoni RF (mobbli) ta’ komunikazzjoni mobbli RF (trasmettituri) u
d-dawl ta' tqaddid kif irrakkomandat hawn taħt, skont il-qawwa massima ta’ ħruġ tat-tagħmir ta' komunikazzjoni.
Distanza ta’ separazzjoni skont il-frekwenza tat-trasmettitur(metri)Qawwa massima stmata tal-ħruġ
ta’ trasmettitur(P f'Watts)

150 kHz – 80 MHz

80 MHz – 800 MHz

800 MHz – 2.5 GHz

0.01

0.12 metri0.035 metru

0.07 metri

0.1

0.37 metri0.11 metri0.22 metri

1

1.7 metri

0.35 metri

0.7 metri

10

3.7 metri1.11 metri

2.22 metri

100

11.7 metri

3.5 metri

7.0 metri

Id-dawl tat-tqaddid ġie ttestjat skont l-IEC 60601-1-2: 2014 u għadda taħt qawwa tal-kamp rradjat ta’ 10 V / m bejn 80-MHz sa 2.5 GHz. Il-valur ta’ 3Vrms
jikkorrispondi għal V1 u l-valur 10 V / m jikkorrispondi għal E1 fil-formuli hawn fuq.
Għal trasmettituri kklassifikati f'qawwa massima ta’ ħruġ mhux elenkata hawn fuq, id-distanza ta' separazzjoni rrakkomandata d f'metri (m) tista’ tiġi stmata billi
tintuża l-ekwazzjoni applikabbli għall-frekwenza tat-trasmettitur, fejn P hija l-ogħla rata ta' qawwa ta’ ħruġ tat-trasmettitur f'watts (\\ t W) skont il-manifattur tattrasmettitur.
NOTA 1 Fi 80 MHz u 800 MHz, tapplika d-distanza ta’ separazzjoni għall-ogħla medda ta' frekwenzi.
NOTA 2 Dawn il-linji gwida jistgħu ma japplikawx fis-sitwazzjonijiet kollha. Il-propagazzjoni elettromanjetika hija affettwata mill-assorbiment u r-riflessjoni minnstrutturi, oġġetti u nies.

201

VALO™ GRANDPLLAMPA POLIMERYZACYJNA
Opis produktuDzięki szerokopasmowemu spektrum lampa utwardzająca VALO Grand jest przeznaczona do utwardzania wszystkich produktów światłoutwardzalnych w zakresie długości fal 385–515 nm
zgodnie z normą ISO 10650. Lampa utwardzająca VALO Grand wykorzystuje akumulatory Ultradent VALO i ładowarkę. Lampę utwardzającą można umieścić w standardowym uchwycie
unitu stomatologicznego lub można ją zamontować na zamówienie za pomocą wspornika dołączonego do zestawu.

Elementy produktu VALO Grand:
•
•
•
•
•
•

1 – Lampa utwardzająca VALO Grand4 – Akumulatory Ultradent VALO
1 – Ładowarka akumulatorów Ultradent VALO z zasilaczem sieciowym 12 V prądu przemiennego klasy medycznej
1 – Zestaw próbki tulei ochronnej VALO
1 – Wielka osłona światła VALO
1 – Wspornik do montażu powierzchniowego światła utwardzającego z podwójną taśmą klejącą

Elementy produktu VALO Grand:Zdjęcie

Przyciski i kontrolki

Szczegółowe instrukcje
Przygotowanie

1) Przed użyciem lampy polimeryzacyjnej należy naładować akumulatory (patrz rozdział Konserwacja akumulatora).
2) Umieścić lampę polimeryzacyjną w standardowym uchwycie montażowym unitu stomatologicznego lub wsporniku montażowym akcesoriów, aż będzie gotowy do użycia.
3) Przed każdym użyciem umieść nowy rękaw barierowy na lampie polimeryzacyjnej.
Higieniczny rękaw barierowy jest dopasowany do lampy polimeryzacyjnej i utrzymuje jej powierzchnię w czystości. Rękaw barierowy pomaga zapobiegać przeniesieniu zakażenia,
ograniczać przyleganie kompozytu stomatologicznego do powierzchni soczewki i lampy polimeryzacyjnej oraz zapobiega przebarwieniom i korozji wywołanej przez roztwory czyszczące.

Uwaga:
LegendaNiski poziom naładowania
akumulatora - wskaźnik
Low Battery - Indicator
Kontrolka trybu/stanu
Przycisk zmiany licznika/
trybu(Górny)W przypadku wszystkich opisanych produktów przed użyciem należy uważnie przeczytać i zrozumieć wszystkie instrukcje i karty charakterystyki.

Wskazania do stosowania/przeznaczenieŹródło oświetlenia do utwardzania fotoaktywowanych materiałów do wypełnień i klejów.

Ostrzeżenia i środki ostrożności
Grupa ryzyka 2
PRZESTROGA: produkt emituje światło UV. W wyniku narażenia może dojść do podrażnienia oczu lub skóry. Należy użyć odpowiedniej osłony.
PRZESTROGA: produkt emituje prawdopodobnie niebezpieczne promieniowanie optyczne. Nie patrzeć na lampę w trakcie pracy. Produkt może mieć szkodliwy wpływ na oczy.
• NIE patrz bezpośrednio w strumień świetlny. Podczas stosowania światła utwardzającego pacjent, lekarz i asystenci powinni zawsze nosić bursztynową ochronę oczu przed
promieniowaniem UV.
• Aby zapobiec ryzyku porażenia prądem, nie wolno modyfikować tego urządzenia. Używaj wyłącznie dołączonego zasilacza i adapterów wtyczek Ultradent VALO. Jeśli te elementy są
uszkodzone, nie używaj i skontaktuj się z działem obsługi klienta Ultradent, aby zamówić wymianę.
• Przenośny sprzęt komunikacyjny RF może pogorszyć wydajność, jeśli jest używany w odległości mniejszej niż 30 cm (12 cali).
• Aby zapobiec nieprawidłowemu działaniu, zwiększonej emisji elektromagnetycznej lub zmniejszonej odporności elektromagnetycznej, należy używać wyłącznie autoryzowanych
akcesoriów, kabli i zasilaczy (patrz sekcja Emisje elektromagnetyczne).
• Aby uniknąć ryzyka pożaru elektrycznego związanego z obsługą akumulatorów:
⸰ NIE WOLNO mieszać akumulatorów z bateriami jednorazowymi lub akumulatorami innego typu.
⸰ NIE próbuj ładować baterii jednorazowych. Ładowanie baterii jednorazowych może skutkować obrażeniami ciała lub uszkodzeniem mienia.
⸰ NIE WOLNO sterylizować w autoklawie ani nie spryskiwać akumulatorów, styków akumulatorów, ładowarki ani zasilacza sieciowego jakimkolwiek płynem. Jeśli na stykachładowarki akumulatora pojawi się korozja, skontaktuj się z Działem Obsługi Klienta firmy Ultradent, aby zamówić wymianę.
⸰ NIE ładuj akumulatorów w pobliżu materiałów łatwopalnych.
⸰ NIE trzymaj ładowarki w gabinecie klinicznym.
• Aby uniknąć ryzyka obrażeń, NIE używaj akumulatorów, które są skorodowane (rdza), wgniecione, wydzielają zapach lub płyn, mają podarte opakowanie, brakuje ich lub są
uszkodzone w inny sposób. Zadzwoń do Działu Obsługi Klienta Ultradent, aby zamówić akumulatory zamienne.
• Używaj wyłącznie zalecanych akumulatorów. Alternatywne akumulatory mogą powodować nieprawidłowe działanie.
• Aby zapobiec ryzyku podrażnienia termicznego lub urazu, należy unikać cykli utwardzania wykonywanych jeden po drugim i nie wystawiać tkanek miękkich jamy ustnej z bliskiej
odległości na dłużej niż 10 sekund w żadnym trybie. Jeśli wymagany jest dłuższy czas utwardzania, należy zastosować wiele cykli utwardzania z przerwami pomiędzy cyklami lub
zastosować produkt o podwójnym utwardzaniu, aby uniknąć nagrzewania tkanek miękkich.
• Należy zachować ostrożność podczas leczenia pacjentów cierpiących na niepożądane reakcje fotobiologiczne lub nadwrażliwość, pacjentów poddawanych chemioterapii lub pacjentów
leczonych lekami fotouczulającymi.
• Urządzenie może być podatne na działanie silnych pól magnetycznych lub statycznych, które mogą zakłócać programowanie. Jeżeli podejrzewasz, że tak się stało, odłącz na chwilę
urządzenie od zasilania, a następnie podłącz je ponownie do gniazdka.
• NIE przecieraj lampy utwardzającej żrącymi lub ściernymi środkami czyszczącymi, nie sterylizuj w autoklawie ani nie zanurzaj w jakiejkolwiek łaźni ultradźwiękowej, środku
202

dezynfekującym, roztworze czyszczącym lub płynie. Niezastosowanie się do dołączonych instrukcji przetwarzania może spowodować, że lampa utwardzająca nie będzie działać.
Aby uniknąć uszkodzenia sprzętu, NIE wkładaj palców, instrumentów ani innych przedmiotów do komory akumulatorów lampy utwardzającej.
Aby uniknąć uszkodzenia sprzętu, NIE próbuj czyścić złotych styków ani żadnej części komory akumulatorów. W razie wątpliwości skontaktuj się z Działem Obsługi Klienta Ultradent.
Aby zapobiec ryzyku zakażenia krzyżowego, tuleje ochronne są przeznaczone do użytku przez jednego pacjenta.
Aby zmniejszyć ryzyko korozji, po użyciu zdjąć tuleję ochronną.
Aby zmniejszyć ryzyko niedostatecznego utwardzenia żywic, nie należy używać lampy utwardzającej, jeśli soczewka jest uszkodzona.

Instalowanie higienicznych rękawów barierowych:

Przycisk(i) zasilania(Górny lub dolny)(Dolny)•••••• Zastosowanie higienicznego rękawa barierowego zmniejsza moc światła o 5–10%. Ze względu na wysoką moc wyjściową lampy polimeryzacyjnej wykazano, że utwardzanie jest
zasadniczo równoważne.
• Lampa polimeryzacyjna musi być czyszczona i odkażana odpowiednimi środkami czyszczącymi i/lub odkażającymi po każdym pacjencie. Patrz punkt Przetwarzanie.

Osłona lampy VALO Cordless:Osłona lampy VALO Cordless jest owalna i może być obracana w celu maksymalnego wykorzystania oraz używana z transparentnym rękawem barierowym.

Użytkowanie

1) Każdy tryb mocy służy do utwardzania materiałów stomatologicznych zawierających fotoinicjatory. Zalecane czasy utwardzania można znaleźć w skróconym przewodniku po trybach.
UWAGA: Lampa polimeryzacyjna jest zaprogramowana tak, aby cyklicznie przełączała się z trybu mocy standardowej przez tryb mocy wysokiej plus do trybu mocy bardzo wysokiej. Na
przykład, aby zmienić tryb mocy standardowej na tryb mocy bardzo wysokiej, konieczne jest przejście do trybu mocy wysokiej plus, a następnie do trybu mocy bardzo wysokiej.
2) Lampa polimeryzacyjna przechowuje ostatnio używany odstęp czasowy i tryb, a domyślnie powraca do tych ustawień po każdej zmianie trybów lub po wyjęciu baterii.

Obsługa
TRYB UTWARDZANIA: Tryb mocy standardowejODSTĘPY CZASOWE: 5, 10, 15, i 20 sekund.
• Lampa polimeryzacyjna przechodzi domyślnie do tego trybu po PIERWSZYM włączeniu zasilania. Kontrolka trybu/stanu będzie zielona, a cztery zielone kontrolki czasu będą sięświecić, wskazując tryb Moc Standardowa.
• Aby zmienić czas, należy szybko nacisnąć przycisk „czas/tryb”.
• Aby rozpocząć utwardzanie, należy nacisnąć przycisk zasilania. Aby przerwać utwardzanie przed upływem odstępu czasowego, należy ponownie nacisnąć przycisk zasilania.

TRYB UTWARDZANIA: Tryb mocy wysokiej plusODSTĘPY CZASOWE: 1, 2, 3, i 4 sekundy.
• W trybie Mocy Standardowej należy nacisnąć i przytrzymać przycisk zmiany czasu/trybu przez 2 sekundy, a następnie zwolnić go. Kontrolka trybu/stanu będzie świecić na
pomarańczowo, a cztery zielone kontrolki czasu będą się świecić i migać, wskazując tryb mocy wysokiej.
• Aby zmienić czas, należy szybko nacisnąć przycisk „czas/tryb”.
• Nacisnąć przycisk zasilania, aby rozpocząć utwardzanie. Aby przerwać utwardzanie przed upływem odstępu czasowego, należy ponownie nacisnąć dowolny przycisk zasilania.
• Aby powrócić do trybu mocy standardowej, należy nacisnąć i przytrzymać przycisk zmiany czasu/trybu przez 2 sekundy, a następnie zwolnić go, co spowoduje przełączenie do trybu
mocy bardzo wysokiej. Należy ponownie nacisnąć i przytrzymać przycisk przez 2 sekundy, a następnie zwolnić go. Kontrolka trybu/stanu będzie zielona, a cztery zielone kontrolki
czasu będą się świecić, wskazując tryb mocy standardowej.

TRYB UTWARDZANIA: Tryb mocy bardzo wysokiejODSTĘP CZASOWY: Tylko 3 sekundy (uwaga: Tryb mocy bardzo wysokiej ma 2-sekundowe opóźnienie bezpieczeństwa na końcu każdego cyklu utwardzania, aby ograniczyć ogrzewanie
podczas kolejnego utwardzania. Po zakończeniu okresu opóźnienia sygnał dźwiękowy wskazuje, że urządzenie jest gotowe do dalszego użycia).
• W trybie mocy standardowej należy nacisnąć przycisk zmiany czasu/trybu przez 2 sekundy, zwolnić go, a następnie nacisnąć go i przytrzymać ponownie przez 2 sekundy i wreszcie
zwolnić go. Kontrolka trybu/stanu będzie świecić na pomarańczowo i migać, a trzy zielone kontrolki czasu będą się świecić i migać, wskazując tryb mocy bardzo wysokiej.
• Aby rozpocząć utwardzanie, należy nacisnąć przycisk zasilania. Aby przerwać utwardzanie przed upływem odstępu czasowego, należy ponownie nacisnąć przycisk zasilania.
• Aby powrócić do trybu mocy standardowej, należy nacisnąć i przytrzymać przycisk zmiany czasu/trybu przez 2 sekundy, a następnie zwolnić go. Kontrolka trybu/stanu będzie zielona,a zielone kontrolki czasu będą się świecić, wskazując tryb mocy standardowej.

Tryb uśpienia: Lampa polimeryzacyjna przechodzi do trybu uśpienia po 60 sekundach bezczynności. Wskazuje na to wolno migająca kontrolka trybu/stanu. Podniesienie lub dotknięcie
urządzenia powoduje aktywację lampy polimeryzacyjnej i jej automatyczny powrót do ostatnio używanego ustawienia. Aby zmaksymalizować żywotność akumulatora, nie należy dotykać
lampy polimeryzacyjnej, gdy nie jest używana.

Sprzątanie1) Po każdym pacjencie należy wyrzucić używane rękawy barierowe ze zwykłymi odpadami.
2) Patrz punkt Przetwarzanie.

Instrukcje dotyczące uchwytów montażowych

1) Uchwyt należy zamontować na płaskiej, niepokrytej olejem powierzchni.
2) Oczyść powierzchnię, przecierając ją alkoholem.
3) Odkleić podkład z taśmy samoprzylepnej na uchwycie.
4) Ustawić uchwyt tak, aby lampa polimeryzacyjna uniosła się do góry po zdjęciu. Docisnąć mocno.

203

Skrócony przewodnik po trybach:
Tryb mocy standardowej
1000 mW/cm2

Tryb

Tryb mocy wysokiej plus
1600 mW/cm2

Tryb mocy bardzo wysokiej
3200 mW/cm2

Przycisk zasilania
Diody LED trybu/czasu

Przyciski czasu

Opcje czasu

5s

10s

15s

20s

1s

2s

3s

4s

Tylko 3s

Aby zmienić czas

Krótko naciśnij i zwolnij przycisk czasu, aby przełączać opcje czasu.

Aby zmienić tryb

Naciśnij i przytrzymaj przycisk czasu przez 2 sekundy, a następnie zwolnij go. Lampa polimeryzacyjna przejdzie do następnego
trybu.

LegendaŚwięcące na stałe diody LED

Migające diody LED

Skrócony przewodnik dotyczący utwardzania:
Tryb utwardzania

Moc* (mW)Natężenie promieniowania*(mW/cm2)

Całkowity czas ekspozycji (sekundy)

Energia** (Dżule)Standard

970

900

5/10/15/20

4,8J/9,7J/14,5J/19,4J

Moc wysoka plus

1 620

1 500

1/2/3/4

1,6J/3,2J/4,8J/6,5J

2 260

2 100

3

6,8J

Dodatkowa*Nominalna moc wyjściowa w odległości 2 mm od czubka soczewki do górnej powierzchni kompozytu.
** Wartości w sekcji całkowitej energii na cykl (dżule) zaokrągla się w górę do najbliższej części dziesiątej.

Skrócony przewodnik dotyczący ostrzeżeń:Ostrzeżenie dotyczące poziomu
zasilania
Wymień akumulatory• Niski poziom naładowania
akumulatora: powolne miganie• Wyłącz: 3 sygnały dźwiękowe,
miganie• Uniemożliwia działanie

Ostrzeżenie dotyczące temperatury

Ostrzeżenie dotyczące serwisu

Ostrzeżenie dotyczące LED

Pozostaw do ostygnięcia

Zadzwoń do działu obsługi klienta w
celu naprawy

Zadzwoń do działu obsługi klienta w
celu naprawy• Brak dźwięku
• Miganie, 2 sekundy• Umożliwia działanie

• Ciągłe 3 sygnały dźwiękowe
• Pojedyncze szybkie mignięcie
• Uniemożliwia działanie

• 3 sygnały dźwiękowe
• Powolne miganie• Uniemożliwia działanie

Ogólne czyszczenie osłony świetła:
Wyczyść bezprzewodową osłonę światła VALO za pomocą dowolnego środka do dezynfekcji powierzchni. NIE dokonywać autoklawu.
Naprawa
Konserwacja wykonywana przez użytkownika
1) Użyj tulei ochronnej, aby zapobiec przyleganiu kompozytu dentystycznego do powierzchni soczewki. Jeśli to konieczne, użyj instrumentu dentystycznego z tworzywa sztucznego lub
stali nierdzewnej, aby ostrożnie usunąć przyklejony kompozyt. Nie używaj narzędzi, które mogą uszkodzić obiektyw.
2) Światłomierze znacznie się od siebie różnią i są zaprojektowane dla określonych końcówek i soczewek światłowodowych. Firma Ultradent zaleca rutynowe sprawdzanie wydruku w
trybie mocy standardowej. UWAGA: prawdziwy wynik numeryczny będzie zniekształcony ze względu na niedokładność popularnych światłomierzy i niestandardowego pakietu diod LED
w lampie utwardzającej
Konserwacja akumulatorówŁadowanie i wymiana akumulatorów
Lampa utwardzająca jest dostarczana z 4 akumulatorami litowo-żelazowo-fosforanowymi. Firma Ultradent zaleca używanie wyłącznie akumulatorów VALO.
Jak ładować akumulatory:
1) Umieść ładowarkę i akumulatory w miejscu, w którym nie będą miały kontaktu z płynami lub sprayem dezynfekującym.
2) Włóż akumulatory do ładowarki biegunem dodatnim (+) skierowanym w stronę lampek kontrolnych na ładowarce.
3) Zielone światła wskazują, że akumulatory są gotowe do użycia.
4) Ładowanie akumulatorów zajmie 1-3 godziny. Pozostaw akumulatory w ładowarce do czasu użycia.
UWAGA: Jeśli czerwona lampka na ładowarce nie zmieni koloru na zielony, gdy akumulatory były ładowane dłużej niż trzy godziny, akumulator może ulec zniszczeniu (uszkodzeniu) i niebędzie można go ładować. Wypróbuj nowy akumulator lub zadzwoń do działu obsługi klienta firmy Ultradent, aby zamówić nowy zestaw akumulatorów.
Jak wymienić/włożyć akumulatory:
1) Zdejmij tylną zatyczkę, przekręcając ją w kierunku przeciwnym do ruchu wskazówek zegara o jedną czwartą obrotu.
2) Wyjmij akumulatory.
3) Najpierw włóż nowe akumulatory, stroną dodatnią (+).
4) Załóż ponownie tylną nasadkę, wyrównując i delikatnie popychając, jednocześnie przekręcając w kierunku zgodnym z ruchem wskazówek zegara. Po całkowitym założeniu nasadka
kliknie.
5) Urządzenie jest gotowe do użycia.
Niski poziom akumulatorów: Lampka utwardzania sygnalizuje użytkownikowi, że nadszedł czas na wymianę akumulatorów, gdy lampka wskaźnika niskiego poziomu akumulatora migana czerwono. Jeżeli poziom naładowania akumulatora będzie zbyt niski, rozlegnie się 3 sygnał dźwiękowy ostrzegawczy, a lampa utwardzająca nie pozwoli na dalszą pracę do czasu
ponownego naładowania akumulatorów lub włożenia nowych akumulatorów. (Patrz Szybki przewodnik ostrzegawczy)
Czas ładowania i żywotność akumulatorów: Żywotność całkowicie naładowanego akumulatora w lampie utwardzającej zależy od trybu/interwału czasu, typu akumulatora, intensywności
użytkowania i wydajności diody LED.
•
Zalecany okres ładowania: Naładuj w razie potrzeby lub gdy zaświeci się wskaźnik niskiego poziomu naładowania akumulatorów. Żywotność zależy od sposobuużytkowania.
Dodatkowe akumulatory: Lampa utwardzająca jest dostarczana z 4 akumulatorami.
•
Naprawa producenta
1) Naprawy mogą być wykonywane wyłącznie przez autoryzowany personel serwisowy. Firma Ultradent ma obowiązek udostępnić personelowi serwisowemu dokumentację niezbędną
do wykonania napraw.
2) Wysyłając urządzenia do naprawy, serwisu lub kalibracji, zawsze wyjmuj akumulatory z lampy utwardzającej i ładowarki. Zapakuj oddzielnie akumulatory, ładowarkę, zasilacz i lampę
utwardzającej w pudełku zwrotnym.
3) Akumulatory należy wysyłać zgodnie z lokalnymi przepisami.
Gwarancja
Ultradent Products, Inc. („Ultradent”) gwarantuje, że produkt ten będzie przez okres 5 lat od daty zakupu, pod warunkiem obsługi zgodnie z instrukcją obsługi dołączoną do produktu, (i)był zgodny we wszystkich istotnych aspektach z specyfikacje określone w dokumentacji Ultradent dołączonej do produktu; oraz (ii) być wolne od wad materiałowych i wykonawczych.
Niniejsza ograniczona gwarancja jest niezbywalna i dotyczy wyłącznie pierwotnego nabywcy i nie rozciąga się na kolejnych właścicieli produktu. Niniejsza ograniczona gwarancja nie
obejmuje żadnych innych akcesoriów, takich jak między innymi akumulatory, ładowarki, adaptery lub soczewki adaptacyjne. Niniejsza ograniczona gwarancja traci ważność, jeśli produkt
ulegnie awarii lub zostanie uszkodzony w wyniku zaniedbania, nadużycia, niewłaściwego użycia, wypadku, modyfikacji, manipulacji, modyfikacji lub nieprzestrzegania odpowiednich
instrukcji użytkowania. Tylko dla celów przykładowych, niniejsza gwarancja nie obejmuje produktu, który upadł i uległ uszkodzeniu. Aby kwalifikować się do niniejszej ograniczonej
gwarancji, dowód zakupu (np. paragon lub podobna dokumentacja) musi zostać przedłożony Ultradent wraz z wadliwym produktem.
Wadliwy produkt spełniający warunki gwarancji określone w niniejszym dokumencie zostanie, według wyłącznego uznania Ultradent, naprawiony lub wymieniony. W żadnym przypadku
odpowiedzialność Ultradent za produkt nie przekroczy ceny zakupu zapłaconej przez kupującego. W żadnym wypadku Ultradent nie ponosi odpowiedzialności za jakiekolwiek szkodypośrednie, przypadkowe, przewidywalne, nieprzewidziane, szczególne lub wtórne powstałe w wyniku lub w związku z użytkowaniem tego produktu.

Odkażanie

Ten produkt jest przeznaczony do stosowania z tuleją ochronną. Informacje na temat konserwacji i ogólnego czyszczenia można znaleźć powyżej.

Przechowywanie i usuwanie
Konserwacja

Ogólne czyszczenie lampy utwardzającej
Po każdym użyciu zwilż gazik lub miękką ściereczkę zatwierdzonym środkiem do dezynfekcji powierzchni i wytrzyj powierzchnię oraz soczewkę. Nieautoryzowane środki czyszczące mogąspowodować uszkodzenie lampy utwardzającej.
DOPUSZCZALNE ŚRODKI CZYSZCZĄCE:• 70% alkohol izopropylowy• 70% etanol204

W przypadku przechowywania lampy polimeryzacyjnej przez okresy dłuższe niż 2 tygodnie lub pakowania urządzenia do transportu należy zawsze wyjąć akumulatory. Jeśli akumulatory
zostaną pozostawione w urządzeniu przez dłuższy czas bez ponownego ładowania, mogą stać się niezdatne do użytku lub niemożliwe do ponownego naładowania. Akumulatorów NIE
WOLNO przechowywać w temperaturach powyżej 40°C (104°F) ani w warunkach bezpośredniego nasłonecznienia.
Przechowywanie i transport lampy polimeryzacyjnej:• Temperatura: +10°C do +40°C (+50°F do +104°F)• Wilgotność względna: 10% do 95%• Ciśnienie otoczenia: 500 hPa do 1060 hPa
Podczas utylizacji odpadów elektronicznych (tj. lamp polimeryzacyjnych, ładowarek, baterii i zasilaczy) należy przestrzegać lokalnych wytycznych dotyczących odpadów i recyklingu.
205

Informacje/dane techniczne

Kwestie techniczne
Akcesoria

Efektywne pasma długości fali utwardzania kompozytu:
Informacje o CE

SKUTECZNE KOMPOZYTOWE PASMA UTWARDZAJĄCE SIĘ NA DŁUGOŚCI FALI
(Wykorzystany zakres długości fali: 385–515 nm)
SPECTRALNA MOC PROMIENIOWANIA (mW/nm)

Pozycja
Osłona lampy VALO CordlessŁadowarka VALO

Akumulatory VALO

Efektowne niższe
pasmo długości fali395-415 nm

Efektywne górne
pasmo długości fal
440-480 nm

Dodatkowa
Moc wysoka
Standard

FALA DŁUGOŚĆ (nm)Cecha

Informacja/specyfikacja

SoczewkaŚrednica 11,7 mm

Zakres długości fali

• Zakres użytecznych długości fal: 385–515 nm• Maks. długości fal: 395–415 nm i 440–480 nm
Tabela porównawcza znamionowej egzytancji promienistej
Przyrząd pomiarowy

Tabela natężenia
światła

206‡ Analizator widma Gigahertz
Egzytancja

Moc całkowita

Apertura aparatu

15 mm

15 mm

Tryb mocy
standardowej(±10%)

900 mW/cm²

970 mW

Tryb mocy wysokiej
plus (±10%)1500 mW/cm²

1615 mW

Tryb mocy bardzo
wysokiej (±10%)2100 mW/cm²

2260 mW

Wyjście promieniowania będzie się różnić wzależności od możliwości instrumentu, metody
pomiaru i rozmieszczenia światła.
‡ Wyjście promieniowania jest zgodne z normą
ISO 10650, gdy jest mierzone za pomocą
analizatora widma Gigahertz-Optik.

Lampa polimeryzacyjna VALO Grand

Parametry znamionowe: IEC 60601-1(bezpieczeństwo), IEC 60601-1-2 (EMC)Waga:• Z akumulatorami: 6 oz. (190 gramów)• Bez akumulatorów: 5 oz. (150 gramów)
Zasilacz ładowarki VALO jest zasilaczem II klasy medycznej i zapewnia izolację
od zasilania sieciowego

Zasilacz doładowarkiWyjście – 12 VDC przy 500 mAWejście – 100 VAC do 240 VAC
Zasilacz ładowarki VALO firmy Ultradent z wtykamimiędzynarodowymi, nr kat. 5930

Parametry znamionowe: IEC 60601-1 (bezpieczeństwo)
Długość przewodu – 6 stóp (1,8 metra)
Zasilacz ładowarki VALO jest zasilaczem II klasy medycznej i zapewnia izolację
od zasilania sieciowego

207

Inne informacje

Cecha

Informacja/specyfikacjaŁadowarka VALOInteligentna ładowarka akumulatorów litowo-fosforanowo-żelazowych 3,6 VDC VALO:• Automatyczne wyłączanie po pełnym naładowaniu• Automatyczne wykrywanie wadliwych akumulatorów• Zabezpieczenia: Termiczne, przed przeładowaniem, przeciwzwarciowe, odwrócona biegunowość
⸰ Czerwona dioda LED – ładowanie⸰ Zielona dioda LED – akumulator całkowicie rozładowany lub naładowany
⸰ Dioda LED nie pali się – zwarcie• Czas ładowania: 1–3 godziny
Parametry znamionowe: CE, WEEE

Akumulatory VALO

Możliwość ładowania: bezpieczna chemia, fosforan litowo-żelazowy (LiFePO4) RCR123A• Napięcie robocze: 3,2 V DC, 400 mAH (wymagana watogodzina)
Klasyfikacje: UL 62133-2, CE, RoHS, WEEE
Ogniwa są przystosowane do > 2000 cykli ładowania/rozładowania, zanim osiągną 70% swojej pojemności. Zaleca się wymianę
akumulatorów co 3-5 lat, ale żywotność może się różnić w zależności od użytkowania. W razie potrzeby wymienić.

Warunki pracy

Temperatura: +10°C do +32°C (+50°F do +90°F)
Wilgotność względna: 10% do 95%Ciśnienie otoczenia: 700 hPa do 1060 hPa

Cykl pracy

Lampa polimeryzacyjna jest przeznaczona do krótkotrwałej pracy. W maksymalnej temperaturze otoczenia (32°C) naprzemiennie: 1
minuta pracy cyklicznej, 30 minut wyłączenia (okres chłodzenia).

Rozwiązywanie problemówJeśli sugerowane rozwiązania nie pozwolą usunąć problemu, należy zadzwonić do firmy Ultradent pod numer 800.552.5512. Poza terytorium USA należy zadzwonić dodystrybutora firmy Ultradent lub dilera materiałów stomatologicznych.
ProblemMożliwe rozwiązania

Lampa nie włącza się1) Naciśnij przycisk zmiany czasu/trybu lub przycisk zasilania, aby „wybudzić” urządzenie z trybu oszczędzania energii.
2) Sprawdź czerwony wskaźnik niskiego poziomu naładowania akumulatora, aby określić stan naładowania akumulatora.
3) Sprawdź, czy nowe akumulatory są prawidłowo włożone do urządzenia.
4) Jeśli czerwone i żółte ostrzegawcze kontrolki LED migają, oznacza to, że lampa polimeryzacyjna osiągnęła wewnętrzny
limit bezpiecznej temperatury. Pozostaw lampę polimeryzacyjną do ostygnięcia przez 10 minut lub użyj chłodnego,
wilgotnego ręcznika, aby szybko ochłodzić urządzenie.
5) Jeśli czerwona ostrzegawcza kontrolka LED świeci i emituje dźwięk w sposób ciągły, należy zadzwonić do działu obsługi
klienta firmy Ultradent w celu przeprowadzenia naprawy.

Lampa nie pozostaje włączona przezżądany czas

1) Sprawdź, czy kontrolki trybu i czasu pokazują właściwą wprowadzoną wartość czasu.
2) Sprawdź wskaźnik poziomu naładowania akumulatora, aby uzyskać informacje o stanie naładowania akumulatora.
3) Sprawdź, czy świeże akumulatory są prawidłowo włożone do urządzenia.

Lampa nieprawidłowo utwardza żywice1) Sprawdź, czy na soczewce nie ma pozostałości utwardzonych żywic/kompozytów.
2) Używając pomarańczowych okularów chroniących przed promieniowaniem UV, sprawdź działanie diod LED.
3) Sprawdź poziom mocy światłomierzem. W przypadku użycia światłomierza firma Ultradent zaleca sprawdzenie lampy
polimeryzacyjnej w trybie mocy standardowej. .
UWAGA: Rzeczywista liczbowa wartość mocy będzie przekłamana z powodu niedokładności typowych światłomierzy iniestandardowego pakietu diod LED w lampie polimeryzacyjnej. Światłomierze różnią się znacznie i są przeznaczone do
konkretnych końcówek światłowodów i soczewek.
4) Sprawdź datę ważności żywicy utwardzającej.
5) Upewnij się, że postępujesz zgodnie z prawidłową techniką według zaleceń producenta.

Akumulatory nie ładują się1) Upewnij się, że akumulatory są włożone do ładowarki we właściwej orientacji i pozostaw je do ładowania przez 1–3
godziny.
2) Jeśli czerwone kontrolki na ładowarce nie zapalą się na zielono, zadzwoń do działu obsługi klienta firmy Ultradent, aby
zamówić nowe akumulatory i/lub ładowarkę.
3) Jeśli na ładowarce nie palą się ani zielone ani czerwone kontrolki, zadzwoń do działu obsługi klienta firmy Ultradent, aby
zamówić lub wymienić ładowarkę i/lub zasilacz.

Ładowarka nie ładuje akumulatorów

1) Upewnij się, że ładowarka jest podłączona, a zasilacz jest podłączony do sprawnego gniazdka elektrycznego.
2) Jeśli na ładowarce nie palą się ani zielone ani czerwone kontrolki, zadzwoń do działu obsługi klienta firmy Ultradent, aby
zamówić nową ładowarkę i/lub zasilacz.

Nie można zmienić trybu ani odstępów
czasowych

1) Przytrzymaj przyciski zmiany czasu/trybu i zasilania jednocześnie do momentu, kiedy seria sygnałów dźwiękowychwskaże, że lampa polimeryzacyjna jest odblokowana.

Wytyczne i deklaracja producenta w sprawie emisji elektromagnetycznych
Lampa polimeryzacyjna jest przeznaczona do stosowania w środowisku elektromagnetycznym określonym poniżej. Klient lub użytkownik powinien upewnić się, że jest
ona używana w takim środowisku.
Test emisji

Zgodność

Środowisko elektromagnetyczne – wytyczne

Emisje RF
CISPR 11

Grupa 1

Lampa polimeryzacyjna wykorzystuje akumulatory i nie podlega działaniom zakłóceń
elektromagnetycznych, fal radiowych ani tłumienia przepięć.

Emisje RF
CISPR 11

Klasa B

Lampa polimeryzacyjna wykorzystuje energię elektryczną i elektromagnetyczną tylko na potrzeby
funkcji wewnętrznych. Dlatego emisje RF są bardzo niskie i prawdopodobnie nie spowodują
zakłóceń w działaniu sprzętu elektronicznego znajdującego się w pobliżu.

Emisje harmoniczne
IEC 61000-3-2

Nie dotyczy

Testy emisji harmonicznych i wahań napięcia nie mają zastosowania w przypadku lampy
polimeryzacyjnej, ponieważ jest ona zasilana za pomocą akumulatorów.

Wahania napięcia/emisje
migotania IEC 61000-3-3

Nie dotyczyLampa polimeryzacyjna może być używana we wszystkich obiektach, w tym mieszkalnych ipodłączonych do publicznej sieci elektroenergetycznej niskiego napięcia zasilającej budynki
mieszkalne.

Wytyczne i deklaracja producenta w sprawie odporności elektromagnetycznej
Lampa polimeryzacyjna jest przeznaczona do stosowania w środowisku elektromagnetycznym określonym poniżej. Klient lub użytkownik powinien upewnić się, że jest
ona używana w takim środowisku.
Test ODPORNOŚCI

Poziom testu IEC 60601

Poziom zgodności

Środowisko elektromagnetyczne – wytyczne

Wyładowanie elektrostatyczne(ESD)±8 kV, styk
±15 kV, powietrze±8 kV, styk
±15 kV, powietrzeŚrodowisko fizyczne powinno być ograniczone donastępujących elementów:
1) Kod IP: IP20
2) Nie zanurzać w płynach.
3) Nie stosować w pobliżu łatwopalnych gazów.
Urządzenie nie jest przeznaczone do stosowania
w obecności środków znieczulających.
4) Zakres wilgotności podczas przechowywania:
10–95%
5) Zakres temperatur podczas przechowywania:
10–40°C

Szybkozmienne zakłócenia
impulsowe
IEC 61000-4-4±2 kV dla linii zasilania
elektrycznego±1 kV dla linii wejścia/wyjścia

±2 kV dla linii zasilania
elektrycznego
Uwaga 1: Lampa polimeryzacyjna
nie ma portów wej./wyj.

Lampa polimeryzacyjna jest zasilana z akumulatorów inie można podłączyć jej do sieci elektrycznej.

Przepięcia
IEC 61000-4-5±1 kV linia do linii±2 kV linia do ziemi±1 kV linia do linii±2 kV linia do ziemi

Spadki, krótkie zaniki i wahanianapięcia na przewodach zasilania<5% U(>95% spadek U przez 0,5 cyklu)<5% U(>95% spadek U przez 0,5 cyklu)

IEC 61000-4-11

40% U(60% spadek U przez 5 cykli)

40% U(60% spadek U przez 5 cykli)

70% U(30% spadek U przez 25 cykli)

70% U(30% spadek U przez 25 cykli)<5% U(>95% spadek U przez 5 s)

<5% U(>95% spadek U przez 5 s)
Uwaga 2: Urządzenie
automatycznie przywraca
ustawienia

IEC 61000-4-2

30 A/mPole magnetyczne
o częstotliwości sieci energetycznej(50/60 Hz)

30 A/mIEC 61000-4-8

Ponieważ lampa polimeryzacyjna jest zasilana z
akumulatorów, nie jest narażona na przemijające skokinapięcia, spadki napięcia, zwarcia, przerwy wahania
mocy z sieci elektrycznej.
Akcesoryjna ładowarka akumulatorów może podlegać
wymienionym poniżej czynnikom, ale jest urządzeniemodrębnym i nie ma krytycznego znaczenia dla działania
lampy polimeryzacyjnej VALO Cordless.
Jeśli napięcie akumulatora lampy polimeryzacyjnej
spadnie do 4 VDC, urządzenie nie zezwoli na jegoużytkowanie. Lampa polimeryzacyjna VALO Cordlesswyłączy się. Po włożeniu nowych akumulatorów i
przywróceniu odpowiednich poziomów mocy lampa
polimeryzacyjna uruchomi się ponownie i powróci dostanu, w jakim znajdowała się przed utratą zasilania.
W przypadku awarii zasilania lampa polimeryzacyjna
automatycznie przywróci ustawienia.
Pola magnetyczne o częstotliwości sieci powinny
być na poziomach charakterystycznych dla typowej
lokalizacji w typowym środowisku mieszkalnym, opieki
domowej, komercyjnym, szpitalnym lub wojskowym.

Uwaga 1: Lampa polimeryzacyjna nie jest wyposażona w porty ani dostępne linie wej./wyj.
208

209

Wytyczne i deklaracja producenta w sprawie odporności elektromagnetycznej systemów nieprzeznaczonych do podtrzymywania życia

Wytyczne i deklaracja producenta dotycząca zalecanych odległości między przenośnymi i mobilnymi urządzeniami komunikacyjnymi RF a lampą VALO CordlessLampa polimeryzacyjna jest przeznaczona do stosowania w środowisku elektromagnetycznym określonym poniżej. Klient lub użytkownik powinien upewnić się, że jest
ona używana w takim środowisku.

Lampa polimeryzacyjna jest przeznaczona do użytkowania w środowisku o kontrolowanych wypromieniowanych zakłóceniach o częstotliwości radiowej. Użytkownik
lampy polimeryzacyjnej może ograniczyć zakłócenia elektromagnetyczne przez utrzymanie zalecanej poniżej minimalnej odległości między przenośnymi i mobilnymi
radiowymi urządzeniami komunikacyjnymi (nadajnikami) a lampą VALO, w zależności od maksymalnej mocy wyjściowej urządzenia komunikacyjnego.

Test ODPORNOŚCI

Poziom testu IEC
60601

Poziom zgodności

Przewodzone
zakłócenia RF

3 Vrms3 VrmsIEC 61000-4-6

150 kHz do 80 MHz

150 kHz do 80 MHzŚrodowisko elektromagnetyczne – wytyczne
Radiostacje przenośne nie powinny być używane w pobliżu lampy polimeryzacyjnej oraz jej
przewodów w odległościach mniejszych niż zalecana odległość ochronna, wyliczona według
równania dla częstotliwości roboczej nadajników:

80 MHz do 800 MHz

800 MHz do 2,5 GHz
Promieniowane
zakłócenia RF

3 V/m

3 V/m

P to maksymalna moc wyjściowa nadajnika w watach (W) podana przez producenta, a d tozalecana odległość w metrach (m).
Natężenie pola stacjonarnych nadajników radiowych, ustalone na podstawie pomiarów
elektromagnetycznych w tereniea powinno być mniejsze niż poziom zgodności w każdym
zakresie częstotliwościb.

IEC 61000-4-3

80 MHz do 2,5 GHz

80 MHz do 2,5 GHzInterference may occur in the vicinity of equipment marked with the following symbol:

Maksymalna znamionowa mocwyjściowa
nadajnika(P w watach)

Odległość w zależności od częstotliwości nadajnika(m)
150 kHz – 80 MHz

80 MHz – 800 MHz

800 MHz – 2,5 GHz

0,01

0,12 m0,035 m0,07 m0,10,37 m0,11 m0,22 m1

1,7 m

0,35 m

0,7 m

10

3,7 m

1,11 m2,22 m100

11,7 m

3,5 m

7,0 m

Lampa polimeryzacyjna została przetestowana zgodnie z normą IEC 60601-1-2: 2014 i przeszła próbę przy natężeniu pola promieniowania 10 V/m w zakresie od 80
MHz do 2,5 GHz. Wartość 3 Vrms odpowiada V1, a wartość 10 V/m odpowiada E1 we wzorach powyżej.
W przypadku nadajników wykorzystujących maksymalną znamionową moc wyjściową niewyszczególnionych powyżej, zalecana odległość oddzielająca d podawana
w metrach (m) może zostać oszacowana z użyciem równania odpowiedniego dla częstotliwości nadajnika, gdzie P oznacza maksymalną znamionową moc wyjściową
nadajnika określaną w watach (W), przy uwzględnieniu danych producenta nadajnika.
UWAGA 1: przy 80 MHz i 800 MHz zastosowanie ma odległość dla wyższego zakresu częstotliwości.
UWAGA 2: podane wskazania mogą nie mieć zastosowania we wszystkich sytuacjach. Propagacja fal elektromagnetycznych jest zależna od ich absorpcji i odbić odkonstrukcji, obiektów oraz ludzi.

UWAGA1: przy 80 MHz i 800 MHz zastosowanie ma wyższy zakres częstotliwości.
UWAGA 2: podane wskazania mogą nie mieć zastosowania we wszystkich sytuacjach. Propagacja fal elektromagnetycznych jest zależna od ich absorpcji i odbić odkonstrukcji, obiektów oraz ludzi.
aDokładnej mocy pola stałych nadajników, takich jak stacje bazowe telefonów radiowych (komórkowych/bezprzewodowych) i przenośnych naziemnych urządzeń
radiowych, radia amatorskie, odbiorniki radiowe AM/FM i odbiorniki TV, nie można przewidzieć teoretycznie. Aby oszacować warunki elektromagnetyczne wynikające
z działania nadajników radiowych, należy przeprowadzić badanie na miejscu. Jeśli natężenie pola zmierzone w miejscu użytkowania lampy polimeryzacyjnej przekraczapowyższe poziomy zgodności, należy obserwować lampę VALO Cordless pod względem prawidłowego działania. Jeśli zaobserwuje się nietypowe objawy, koniecznemogą być dodatkowe działania, takie jak zmiana ustawienia lub orientacji lampy polimeryzacyjnej.
b
W przypadku zakresu częstotliwości od 150 kHz do 80 MHz moc pola powinna być mniejsza niż 3 V/m.

210

211

VALO™ GRAND

ROLAMPĂ DE POLIMERIZARE
Descrierea produsului

Cu spectrul său de bandă largă, lumina de polimerizare VALO Grand este concepută pentru a polimeriza toate produsele fotopolimerizate în intervalul de lungimi de undă de 385-515
nm conform ISO 10650. Lumina de polimerizare VALO Grand utilizează bateriile reîncărcabile Ultradent VALO și încărcătorul de baterii. Lumina de polimerizare este concepută pentru a fiașezată într-un suport standard pentru unitatea dentară sau poate fi montată la comandă cu ajutorul suportului de montare pe suprafață.

Componentele produsului VALO Mare:
•
•
•
•
•
•

1 – Lumină de polimerizare VALO Grand4 – Baterii reîncărcabile Ultradent VALO
1 – Încărcător de baterii Ultradent VALO cu sursă de alimentare AC de 12VDC de calitate medicală
1 – Pachet cu mostre pentru manșoane barieră VALO
1 – Scut luminos VALO Grand1 – Suport de montare pe suprafață cu bandă adezivă cu lipire dublă

Prezentare generală a comenzilor:

Fotografie

1) Încărcați bateriile înainte de a utiliza lumina de polimerizare (consultați secțiunea Întreținerea bateriilor).
2) Înainte de fiecare utilizare, așezați un manșon de barieră nou peste lumina de polimerizare, reducând la minimum ridurile de pe lentilă pentru rezultate optime.
• Pentru a preveni contaminarea încrucișată și pentru a împiedica materialul compozit dentar să adere la suprafața lentilelor și a corpului dispozitivului, trebuie utilizat un manșon debarieră aprobat de Ultradent peste lumina de polimerizare VALO la fiecare utilizare. Manșoanele de barieră sunt destinate utilizării pentru un singur pacient.

Ecran de protecție împotriva luminii VALO fără fir:

Ecranul de protecție împotriva luminii VALO fără fir are formă ovală, poate fi rotit pentru o utilizare maximă și poate fi utilizat cu o husă de protecție transparentă.

Utilizare

Butoane și lumini

Legendă

Lumini pentru temporizare
Nivel scăzut baterie - Indicator
Bec mod/stare
Buton schimbare mod/temporizare(parte superioară)Pentru toate produsele descrise, citiți cu atenție și înțelegeți toate instrucțiunile și informațiile din fișa cu date de securitate înainte de utilizare.

Instrucțiuni de utilizare/scop preconizatSursă de iluminare pentru polimerizarea materialelor dentare pentru restaurare și a adezivilor fotopolimerizabili.

Avertismente și precauții
Grupul de risc 2ATENȚIONARE UV emise de acest produs. Expunerea poate cauza iritații ale pielii sau ochilor. Utilizați protecție adecvată.
ATENȚIE Acest produs poate emite radiații optice potențial periculoase. Nu priviți direct lampa. Aceasta poate fi periculos pentru ochi.
• NU priviți direct în lumina emisă. Pacientul, clinicianul și asistenții trebuie să folosească întotdeauna protecție oculară de culoare chihlimbar pe durata utilizării luminii de polimerizare.
• Pentru a preveni riscul de electrocutare, nu este permisă modificarea acestui echipament. Utilizați numai sursa de alimentare luminoasă Ultradent VALO inclusă și adaptoarele depriză. În cazul în care aceste componente sunt deteriorate, nu le utilizați și apelați Serviciul Clienți Ultradent pentru a comanda un înlocuitor.
• Echipamentele portabile de comunicații RF pot reduce performanțele dacă sunt utilizate la o distanță mai mică de 30 cm (12 in.).
• Utilizați numai accesorii, cabluri și surse de alimentare autorizate pentru a preveni funcționarea necorespunzătoare, creșterea emisiilor electromagnetice sau scăderea rezistenței
electromagnetice (consultați secțiunea Emisii electromagnetice).
• Pentru a evita riscul de incendiu electric asociat cu manipularea bateriilor:
⸰ NU amestecați bateriile reîncărcabile cu baterii nereîncărcabile sau cu alte tipuri de baterii.
⸰ NU încercați să încărcați baterii nereîncărcabile. Încărcarea bateriilor nereîncărcabile poate duce la rănire sau la daune materiale.
⸰ NU sterilizați în autoclavă și NU stropiți bateria, contactele bateriei, încărcătorul sau sursa de alimentare de curent alternativ cu niciun fel de lichid. Dacă apare coroziune pe
contactele încărcătorului, apelați Serviciul Clienți Ultradent pentru a comanda o piesă de schimb.
⸰ NU încărcați bateriile în apropierea materialelor inflamabile.
⸰ NU păstrați încărcătorul în cabinetul operator clinic.
• Pentru a evita riscul de rănire, NU folosiți baterii care sunt corodate (rugină), îndoite, care emit un miros sau fluide, care au ambalajul rupt sau lipsește sau care sunt deteriorate în alt
mod. Sunați la Serviciul Clienți Ultradent pentru a comanda baterii de schimb.
• Utilizați numai bateriile recomandate. Bateriile alternative pot provoca defecțiuni.
• Pentru a preveni riscul de iritație termică sau de rănire, evitați ciclurile de polimerizare succesive și nu expuneți țesuturile moi orale în imediata apropiere pentru mai mult de 10
secunde în orice mod. Dacă este necesar un timp de polimerizare mai lung, utilizați mai multe cicluri de polimerizare cu perioade de repaus între cicluri sau utilizați un produs cu
polimerizare dublă pentru a evita încălzirea țesuturilor moi.
• Fiți precauți atunci când tratați pacienții care prezintă reacții fotobiologice adverse sau sensibilități, pacienții care urmează un tratament de chimioterapie sau pacienții care sunt tratați
cu medicamente fotosensibilizante.
• Acest aparat poate fi sensibil la câmpuri magnetice puternice sau câmpuri electrice statice, ceea ce ar putea afecta programarea. Dacă bănuiți că acest lucru s-a întâmplat,
deconectați momentan dispozitivul și apoi reconectați-l la priză.
• NU curățați lumina de polimerizare cu detergenți caustici sau abrazivi, nu o introduceți în autoclavă și nu o scufundați în niciun fel de baie cu ultrasunete, dezinfectant, soluție decurățare sau lichid. Nerespectarea instrucțiunilor de procesare incluse poate face ca lumina de polimerizare să devină nefuncțională.
212

Instrucțiuni etapizatePregătireNotă:

Buton de pornire(parte superioară sau parte inferioară)(parte inferioară)• Pentru a evita deteriorarea echipamentului, NU introduceți degetele, instrumentele sau alte obiecte în compartimentul pentru baterii al lămpii de polimerizare.
• Pentru a evita deteriorarea echipamentului, NU încercați să curățați contactele electrice sau orice parte a compartimentului bateriei. Sunați Serviciul Clienți Ultradent dacă există oproblemă.
• Pentru a preveni riscul de contaminare încrucișată, manșoanele barieră sunt de unică folosință pentru pacienți.
• Pentru a reduce riscul de coroziune, îndepărtați manșonul barieră după utilizare.
• Pentru a reduce riscul unor rășini insuficient polimerizate, nu utilizați lumina de polimerizare dacă lentila este deteriorată

1) Fiecare mod de alimentare este utilizat pentru polimerizarea materialelor dentare cu inițiatori foto. Consultați Ghidul rapid pentru moduri pentru perioadele recomandate depolimerizare.
NOTĂ: Lampa de polimerizare este programată să schimbe ciclul în mod succesiv de la modul de Putere standard la Putere ridicată plus la modul de Putere Xtra. De exemplu, pentru a
trece de la modul Putere standard la modul Putere Xtra, este necesar să treceți în modul Putere ridicată plus și apoi la modul Putere Xtra.
2) Lampa de polimerizare memorează cel mai recent interval de temporizare și mod de funcționare utilizate și va reveni la acestea la fiecare schimbare a modurilor sau dacă bateriile
sunt îndepărtate.

OperareMOD DE POLIMERIZARE: Modul de Putere standard

INTERVALE DE TEMPORIZARE: 5, 10, 15, și 20 de secunde.
• Acest mod este setat implicit la PRIMA pornire a lămpii de polimerizare. Becul de mod/stare luminează în verde, iar cele patru becuri de temporizare verzi luminează, indicând modul
de Putere standard.
• Pentru a schimba intervalele de temporizare, apăsați rapid butonul mod/timp.
• Apăsați butonul de pornire pentru realiza polimerizarea. Pentru a opri polimerizarea înainte de finalizarea unui interval de temporizare, apăsați din nou butonul de pornire.

MOD DE POLIMERIZARE: Modul Putere ridicată plus

INTERVALE DE TEMPORIZARE: 1, 2, 3, și4 de secunde.
• Din modul Putere standard, apăsați și țineți apăsat butonul de schimbare timp/mod timp de 2 secunde și eliberați-l. Becul de mod/stare va lumina portocaliu, iar cele patru becuri
verzi de temporizare luminează și clipesc, indicând modul Putere ridicată.
• Pentru a schimba intervalele de temporizare, apăsați rapid butonul mod/timp.
• Apăsați butonul de pornire pentru a realiza polimerizarea. Pentru a opri polimerizarea înainte de finalizarea unui interval de temporizare, apăsați din nou butonul de pornire.
• Pentru a reveni la modul Putere standard, apăsați și țineți apăsat butonul de schimbare timp/mod timp de 2 secunde și eliberați-l, prin aceasta putând trece la modul Putere Xtra.
Apăsați și mențineți apăsat din nou timp de 2 secunde și eliberați. Becul de mod/stare va lumina verde, iar cele patru becuri de temporizare verzi luminează, indicând modul dePutere standard.

MOD DE POLIMERIZARE: Modul Putere Xtra

INTERVAL DE TEMPORIZARE: Numai 3 secunde (Notă: Modul Putere Xtra are o întârziere de siguranță de 2 secunde la sfârșitul fiecărui ciclu de polimerizare pentru a limita încălzirea întimpul polimerizării consecutive. La sfârșitul întârzierii, semnalul sonor indică faptul că unitatea este pregătită pentru utilizare continuă).
• Din modul Putere standard, apăsați butonul de schimbare timp/mod timp de 2 secunde, eliberați, țineți apăsat din nou timp de 2 secunde și eliberați. Becul de mod/stare va lumina
portocaliu și va clipi, iar trei dintre becurile de temporizare verzi vor lumina și vor clipi, indicând modul Putere Xtra.
• Apăsați butonul de pornire pentru realiza polimerizarea. Pentru a opri polimerizarea înainte de finalizarea unui interval de temporizare, apăsați din nou butonul de pornire.
• Pentru a reveni la modul Putere standard, apăsați și țineți apăsat butonul timp/mod timp de 2 secunde și eliberați-l. Becul de mod/stare luminează în verde, iar becurile detemporizare verzi luminează, indicând modul de Putere standard.

Modul de veghe: Lampa de polimerizare va intra în modul de veghe după 60 de secunde de inactivitate, indicat printr-o clipire lentă a becului mod/stare. Ridicarea sau atingerea
aparatului va porni lampa de polimerizare și va reveni automat la ultima setare utilizată. Pentru a maximiza durata de viață a bateriei, lăsați lămpile de tratare neperturbate atunci cândnu le folosiți.

Curățare1) Aruncați husele de protecție utilizate cu deșeurile standard după fiecare pacient.
2) Consultați secțiunea Prelucrare.

Instrucțiuni de montare a suportului

1) Suportul trebuie montat pe o suprafață plană, fără ulei.
2) Curățați suprafața cu spirt.
3) Îndepărtați partea posterioară a benzii adezive a suportului.
4) Poziționați suportul astfel încât lămpile de polimerizare să se ridice în sus atunci când sunt scoase. Apăsați ferm pe poziție.

Ghidul rapid pentru moduri:
Mod

Putere standard
1000 mW/cm2

Putere ridicată plus
1600 mW/cm2

Putere Xtra
3200 mW/cm2

213

Întreținerea baterieiÎncărcarea și schimbarea bateriilor
Lumina de polimerizare este livrată cu 4 baterii reîncărcabile litiu-fosfat de fier. Ultradent recomandă utilizarea exclusivă a bateriilor reîncărcabile VALO.

Buton de pornire
LED-uri Mod/timp

Butoane pentru timpOpțiuni de timp

5s

10s

15s

20s

1s

2s

3s

4s

Numai 3s

Pentru a modifica timpulApăsați și eliberați rapid butonul de timp pentru a parcurge opțiunile de timp.

Modificarea modurilorȚineți apăsat butonul pentru timp pentru 2 secunde și eliberați-l. Lampa de polimerizare va trece în modul următor.

Legendă

LED-uri care luminează constantLED-uri care clipesc

Ghid rapid de polimerizare:
Mod de polimerizare

Putere* (mW)

Iradiere* (mW/cm2)

Timp total de expunere (secunde)

Energie** (Jouli)

Standard

970

900

5/10/15/20

4.8J/9.7J/14.5J/19.4J

High Power Plus (Puteremare Plus)

1,620

1,500

1/2/3/4

1.6J/3.2J/4.8J/6.5J

Xtra

2,260

2,100

3

6.8J

Ghid rapid de avertismente:*Ieșire nominală la o distanță de 2 mm de la vârful lentilei la suprafața superioară a compozitului.
**Valorile din secțiunea energie totală pe ciclu (jouli) sunt rotunjite la cea mai apropiată zecime.

Avertisment nivel de putereÎnlocuiți bateriile• Nivel scăzut baterie: clipire lentă
• Oprire: 3 semnale sonore, clipire• Împiedică funcționarea

Avertisment temperatură

Avertisment pentru service

Avertisment LED

Permiteți răcirea unității

Apelați Serviciul de relații cu clienții
pentru reparații

Apelați Serviciul de relații cu clienții
pentru reparații

• Nici un sunet• Clipește la fiecare 2 secunde• Permite funcționarea• 3 semnale sonore continue• O singură clipire rapidă
• Împiedică funcționarea• 3 semnale sonore• Clipește lent• Împiedică funcționarea

Cum să încărcați bateriile:1) Așezați încărcătorul și bateriile într-o zonă în care nu vor intra în contact cu lichide sau spray-uri dezinfectante.
2) Introduceți bateriile în încărcător cu capătul pozitiv (+) îndreptat spre indicatoarele luminoase de pe încărcător.
3) Luminile verzi indică faptul că bateriile sunt gata de utilizare.
4) Încărcarea bateriilor va dura 1-3 ore. Lăsați bateriile în încărcător până când sunt gata de utilizare.
NOTĂ: Dacă lumina roșie de pe încărcător nu devine verde când bateriile s-au încărcat mai mult de trei ore, este posibil ca bateria să fie afectată ("deteriorată") și să nu poată fi încărcată.
Încercați o baterie nouă sau apelați Serviciul Clienți Ultradent pentru a comanda un set nou de baterii reîncărcabile.
Cum se schimbă/introduc bateriile:
1) Scoateți capacul din spate răsucind în sens invers acelor de ceasornic un sfert de rotație.
2) Scoateți bateriile.
3) Introduceți mai întâi bateriile noi cu partea pozitivă (+).
4) Montați din nou capacul din spate prin aliniere și împingând ușor în timp ce îl rotiți în sensul acelor de ceasornic. Capacul va face clic când este complet atașat.
5) Unitatea este gata de utilizare.
Baterii descărcate: Indicatorul luminos de polimerizare semnalează utilizatorului că este timpul să schimbe bateriile atunci când indicatorul luminos al bateriei descărcate clipește roșu.
În cazul în care încărcătura bateriei devine prea mică, se va auzi un semnal sonor de avertizare cu 3 semnale sonore, iar lumina de polimerizare nu va permite continuarea funcționării
până când bateriile nu sunt reîncărcate sau până când nu sunt introduse baterii noi. (Consultați Ghidul de avertizare rapidă)Timpul de încărcare și durata de viață a bateriilor: Durata de viață a bateriei complet încărcate în lumina de polimerizare depinde de intervalul Mod/Timp, de tipul de baterie, decantitatea de utilizare și de eficiența LED-ului.
• Interval de reîncărcare recomandat: Reîncărcați după cum este necesar sau, atunci când se aprinde indicatorul de baterie descărcată. Durata de viață depinde de utilizare.
• Baterii suplimentare: Lumina de polimerizare este livrată cu 4 baterii reîncărcabile.
Reparații efectuate de producător
1) Reparațiile trebuie efectuate numai de către personalul de service autorizat. Ultradent trebuie să furnizeze personalului de service documentația pentru efectuarea reparațiilor.
2) Atunci când trimiteți aparatele pentru reparații, service sau calibrări, scoateți întotdeauna bateriile de la lumina de polimerizare și de la încărcător. Împachetați separat bateriile,încărcătorul, adaptorul și lumina de polimerizare în cutia de retur.
3) Expediați bateriile în conformitate cu reglementările locale.
Garanție
Ultradent Products, Inc. ("Ultradent") garantează că acest produs, pentru o perioadă de 5 ani de la data achiziționării, atunci când este utilizat în conformitate cu instrucțiunile deutilizare incluse cu produsul, (i) este conform în toate aspectele materiale cu specificațiile stabilite în documentația Ultradent care însoțește produsul; și (ii) este lipsit de defecte dematerial și de manoperă.
Această garanție limitată nu este transmisibilă și se aplică numai cumpărătorului inițial și nu se extinde la proprietarii ulteriori ai produsului. Această garanție limitată nu acoperă nicioaltă componentă accesorie, cum ar fi, dar fără a se limita la, baterii, încărcătoare sau lentile adaptive. Această garanție limitată este nulă în cazul în care produsul nu funcționează sau
este deteriorat din cauza neglijenței, abuzului, utilizării necorespunzătoare, accidentului, modificării, manipulării, alterării sau nerespectării instrucțiunilor de utilizare aplicabile. Doarca exemplu, un produs care este scăpat și deteriorat nu este acoperit de această garanție. Pentru a beneficia de această garanție limitată, trebuie trimisă la Ultradent o dovadă decumpărare (de exemplu, chitanța de vânzare sau un document similar) împreună cu produsul defect.
Un produs defect care îndeplinește condițiile de garanție stabilite în prezentul document va fi, la discreția exclusivă a Ultradent, fie reparat, fie înlocuit. În niciun caz răspunderea
Ultradent pentru produs nu va depăși prețul de achiziție plătit de cumpărător. În niciun caz, Ultradent nu va fi răspunzătoare pentru daune indirecte, accidentale, prevăzute, neprevăzute,speciale sau de consecință care rezultă din sau în legătură cu utilizarea acestui produs.

Prelucrare

Acest produs este destinat utilizării cu un manșon de barieră. Consultați mai sus pentru întreținere și curățare generală.

Depozitare și eliminareÎntreținereCurățarea generală a luminii de polimerizareDupă fiecare utilizare, umeziți un tifon sau o cârpă moale cu un dezinfectant de suprafață aprobat și ștergeți suprafața și lentila. Produsele de curățare neautorizate pot provoca
deteriorarea luminii de polimerizare.
PRODUSE DE CURĂȚARE ACCEPTABILE:• 70% alcool izopropilic• 70% etanolCurățarea generală a luminii de polimerizareCurățați ecranul luminos VALO Cordless folosind orice dezinfectant de suprafață. NU folosiți autoclavarea.
Reparații
Întreținerea efectuată de utilizator
1) Folosiți un manșon de barieră pentru a preveni aderența compozitului dentar la suprafața lentilei. Dacă este necesar, utilizați un instrument dentar din plastic sau din oțel inoxidabilpentru a îndepărta cu grijă orice compozit aderat. Nu utilizați instrumente care vor deteriora lentila.
2) Aparatele de măsurat lumina diferă foarte mult și sunt concepute pentru vârfuri și lentile de ghidare a luminii specifice. Ultradent recomandă verificarea de rutină a randamentului în
modul Standard Power. NOTĂ: ieșirea numerică reală va fi deviată din cauza inexactității aparatelor de măsură a luminii obișnuite și a pachetului de LED-uri personalizat din lumina depolimerizare.
214Dacă depozitați lampa de polimerizare pentru perioade mai lungi de 2 săptămâni sau când o împachetați pentru a călători, scoateți întotdeauna bateriile. Dacă bateriile sunt lăsate în
aparat pentru perioade lungi de timp fără reîncărcare, ele pot deveni nefuncționale sau nu se mai pot încărca. Nu depozitați bateriile la temperaturi de peste 40 °C (104 °F) sau în lumina
directă a soarelui.
Depozitarea și transportul lămpii de polimerizare:• Temperatura: de la + 10 °C până la + 40 °C (+ 50 °F până la + 104 °F)• Umiditate relativă: de la 10% până la 95%• Presiunea ambientală: de la 500 hPa până la 1060 hPa
Atunci când eliminați deșeurile electronice (adică lămpi de polimerizare, încărcătoare, baterii și surse de alimentare), urmați orientările locale privind deșeurile și reciclarea.

Considerente tehnice
Accesorii
Articol

Informații CE

Ecran de protecție împotriva luminii
VALO fără fir
215

ArticolInformații CE

Informații/date tehnice

Benzi efective de lungime de undă pentru polimerizarea compozitului:Încărcător VALO

PUTERE RADIANTĂ SPECTRALĂ (mW/nm)

BENZI DE LUNGIMI DE UNDĂ COMPOZITE EFICIENTE DE ÎNTĂRIRE
(Gama de lungimi de undă utilizabile: 385-515 nm)

Baterii VALO

Eficace mai mic
Banda de lungimi
de undă
395-415 nm

Banda efectivă delungime de undăsuperioară
440-480 nm

XtraHigh Power (Putere mare)Standard

WAVELENGTH (nm)Atribut

Informații/specificații

Lentilă

Diametru 11,7 mm

Spectrul lungimilorde undă

• Spectrul lungimilor de undă utilizat: 385 – 515 nm• Lungimi de undă de vârf: 395 - 415 nm și 440 - 480 nm
Diagrama nominală de comparație a fluxului radiantInstrument demăsurareTabel de intensitate
luminoasă

216‡ Analizor de spectru Gigahertz
Flux

Putere totală

Deschiderea
instrumentului demăsurare

15 mm

15 mm

Putere standard(±10%)

900 mW/cm²

970 mW

Putere ridicată plus(±10%)

1500 mW/cm²

1615 mW

Putere Xtra (±10%)2100 mW/cm²

2260 mW

Exitanța radiantă variază în funcție decapacitatea instrumentului, de metoda demăsurare și de amplasarea luminii.
‡ Exitanța radiantă este conformă cu ISO
10650 atunci când este măsurată cu unanalizor de spectru Gigahertz-Optik.

Lampă depolimerizare VALO
Mare

Clasificări: IEC 60601-1 (Siguranță), IEC 60601-1-2(CEM)

Greutate:• Cu baterii: 6 oz. (190 grame)• Fără baterii: 5 oz. (150 grame)
Dimensiuni: (8 x 1,28 x 1,06) inci, (203 x 32,5 x 27) mm

Încărcătorul dealimentareIeșire - 12 V c.c. la 500 mA
Intrare - de la 100 Va.c. la 240 V a.c.
Încărcător de alimentare Ultradent P/N 5930 VALO
cu mufă internațională

Clasificări: IEC 60601-1 (Siguranță)Lungimea cablului - 6 picioare (1,8 metri)Încărcătorul de alimentare VALO este o sursă de alimentare de clasă medicală
II și este izolat de la alimentarea principală

217

Atribut

Informații/specificații

Încărcător VALO

Încărcător inteligent baterie litiu-fier-fosfat VALO 3,6 V c.c.
• Oprire automată la încărcarea completă
• Detectare automată a bateriilor defecte• Protecții: Termică, supraîncărcare, scurtcircuit, polaritate inversă
⸰ LED roșu – încărcare
⸰ LED verde – baterie epuizată sau complet încărcată
⸰ LED oprit – scurtcircuitt• Timp de încărcare: 1 - 3 ore
Clasificare: CE, DEEE

Informații diverse
Ghid și declarația producătorului cu privire la emisii electromagneticeLampa de polimerizare este destinată utilizării în mediul electromagnetic specificat mai jos. Clientul sau utilizatorul trebuie să se asigure că acesta se utilizează într-un
astfel de mediu.

Baterii VALO

Reîncărcabil: Chimie sigură Litiu-fosfat de fier (LiFePO4) RCR123A
Tensiune de lucru: 3.2VDC, 400mAH (Watt oră necesară)•Valori nominale: UL 62133-2, CE, RoHS, WEEE
Celulele sunt evaluate pentru >2000 de cicluri de încărcare/descărcare înainte de a atinge 70% din capacitate. Se recomandă
înlocuirea bateriilor la fiecare 3-5 ani, dar durata de viață poate varia în funcție de utilizare. înlocuiți dacă este necesar.

Condiții de operare

Temperatura: de la + 10 °C până la + +32°C (+ 50 °F până la + +90°F)
Umiditate relativă: de la 10% până la 95%
Presiunea ambientală: de la 700 hPa până la 1060 hPa

Ciclul de utilizare

Lampa de polimerizare este concepută pentru utilizări cu termen scurt. La temperatura ambientală maximă (32 °C) cicluri consecutive
PORNIT de 1 minut, 30 de minute OPRIRE (perioadă de repaus).

DepanareDacă soluțiile sugerate mai jos nu remediază problema, apelați Ultradent la 800.552.5512. În afara Statelor Unite, apelați distribuitorul Ultradent sau la distribuitorul
de produse dentare.
Problemă

Soluții posibile

Lumina nu se aprinde

1) Apăsați butonul Timp/mod sau butonul de Pornire pentru a ieși din modul economisire energie.
2) Verificați indicatorul roșu de nivel scăzut al bateriei pentru starea încărcării bateriei.
3) Verificați dacă bateriile noi sunt introduse corect în aparat.
4) Dacă LED-urile de avertizare roșii și galbene luminează intermitent, aceasta înseamnă că lampa de polimerizare a atins
limita de siguranță internă a temperaturii. Lăsați ca lampa de polimerizare să se răcească timp de 10 minute sau utilizați
un prosop răcoros umed pentru a răci rapid unitatea.
5) Dacă LED-ul roșu de avertizare clipește și emite un semnal sonor continuu, apelați Serviciul de relații cu clienții Ultradentpentru reparații.

Lumina nu rămâne aprinsă pentru
timpul dorit

1) Verificați luminile Mod și Timp pentru introducerea corectă a timpului.
2) Verificați indicatorul nivelului scăzut al bateriei pentru starea încărcării bateriei.
3) Verificați dacă bateriile noi sunt introduse corect în unitate.

Lumina nu polimerizează adecvatrășinile

1) Verificați dacă există reziduuri de rășină/compozit pe lentilă.
2) Folosind echipament de protecție pentru ochi cu lentile portocalii, verificați dacă lumina LED-ului funcționează.
3) Verificați nivelul de putere cu ajutorul fotometrului. Dacă utilizați un fotometru, Ultradent recomandă verificarea lămpii
de polimerizare în modul Putere standard.
NOTĂ: Ieșirea numerică reală nu va fi prezentată din cauza inexactității fotometrelor obișnuite și a setului LED personalizat
utilizat de lampa de polimerizare. Fotometrele diferă foarte mult și sunt concepute pentru vârfuri specifice de ghidare aluminii și lentile.
4) Verificați data de expirare pe rășina de polimerizare.
5) Asigurați-vă că se respectă tehnica adecvată conform recomandărilor producătorului.

Bateriile nu se încarcă

1) Asigurați-vă că bateriile sunt introduse în încărcător în direcția corectă și lăsați bateriile să se încarce timp de 1-3 ore.
2) Dacă becurile roșii ale încărcătorului nu se schimbă în verde, apelați Serviciul de relații cu clienții Ultradent pentru acomanda baterii de schimb și/sau încărcător.
3) Dacă becurile verzi și cele roșii ale încărcătorului nu sunt vizibile, apelați Serviciul de relații cu clienții Ultradent pentru acomanda sau a înlocui încărcătorul și/sau adaptorul de curent alternativ.

Încărcătorul nu încarcă bateriile

1) Asigurați-vă că încărcătorul este conectat, iar adaptorul de curent alternativ este cuplat la o priză de alimentare carefuncționează.
2) Dacă becurile roșii ale încărcătorului nu se schimbă în verde, apelați Serviciul de relații cu clienții Ultradent pentruîncărcător nou și/sau adaptor de curent alternativ.

Modul sau intervalele de timp nu se
pot schimba1) Țineți butoanele Timp/mod și butoanele de pornire până când o serie de semnale sonore indică deblocarea lămpii depolimerizare.

Test de emisii

Conformitate

Mediul electromagnetic - orientare

Emisii RF
CISPR 11

Grupa 1

Lampa de polimerizare folosește bateriile și nu este afectată de EMI, RF sau suprimarea
supratensiunii.

Emisie RF
CISPR 11

Clasa B

Emisii armonice
IEC 61000-3-2

N/A

Lampa de polimerizare utilizează energie electrică și electromagnetică numai pentru funcțiile
interne. Prin urmare, orice emisii RF sunt foarte scăzute și nu sunt susceptibile de a provoca
interferențe în echipamentele electronice din apropiere.

Volumul fluctuațiilor detensiune/emisiilor de pâlpâire
IEC 61000-3-3

N/A

Testarea pentru emisii armonice și fluctuații de tensiune nu sunt aplicabile lămpii de polimerizare
deoarece este alimentată cu baterii.
Lampa de polimerizare este adecvată pentru utilizare în toate unitățile, inclusiv în unitățile localeși cele care sunt conectate direct la rețeaua publică de alimentare cu energie electrică de joasă
tensiune care alimentează clădirile pentru uz casnic.

Ghid și declarația producătorului cu privire la imunitate electromagnetică
Lampa de polimerizare este destinată utilizării în mediul electromagnetic specificat mai jos. Clientul sau utilizatorul trebuie să se asigure că acesta se utilizează într-un
astfel de mediu.
Testul de imunitate

Nivelul de testare IEC 60601

Nivelul de conformitate

Ghidul mediului electromagneticDescărcare electrostatică (ESD)± 8 kV contact± 15 kV aer± 8 kV contact± 15 kV aer

Mediul fizic ar trebui să se limiteze la următoarele:
1) Codul IP: IP20
2) Nu scufundați în lichid.
3) Nu utilizați lângă gaze inflamabile. Unitatea este
non-APG și non-AP.
4) Interval de umiditate pentru depozitare:
10% - 95%
5) Interval de temperatură pentru depozitare:
10 °C - 40 °C

Trenurile de impulsuri rapide
de tensiune
IEC 61000-4-4± 2 kV pentru liniile de alimentare
cu energie electrică
± 1 kV pentru liniile de intrare/ieșireSupratensiune tranzitorie
IEC 61000-4-5± 1 kV linie la linie± 2 kV la pământ

± 2 kV pentru liniile de alimentare Lampa de polimerizare este alimentată prin baterii
cu energie electrică
și nu se poate conecta la alimentarea principală cu
Nota 1: Lampa de polimerizare nu curent alternativ.
are porturi I/0
Deoarece lampa de polimerizare este alimentată
prin baterii, aceasta nu este supusă impulsurilor de± 1 kV linie la linie
tensiune, supratensiunilor, scurgerilor de tensiune,± 2 kV la pământ
scurtcircuitelor, întreruperilor sau variațiilor de putere
ale alimentării principale cu curent alternativ.
<5% U(>95% cădere în U pentru 0,5Încărcătorul de baterii furnizat ca accesoriu poate facecicluri)obiectul celor indicat mai sus, dar este separat și nu
este vital pentru funcționarea lămpii de polimerizare
40% U(60% cădere în U pentru 5 cicluri) VALO fără fir.

IEC 61000-4-2

Tensiune, căderi, scurtcircuite,<5% Uîntreruperi și variații ale cablurilor (>95% cădere în U pentru 0,5
de intrare ale sursei de alimentare cicluri)
IEC 61000-4-11

40% U(60% cădere în U pentru 5 cicluri)Dacă tensiunea bateriei lămpii de polimerizare scade
70% U
70% U
(30% cădere în U pentru 25 cicluri) (30% cădere în U pentru 25 cicluri) la 4 V c.c., unitatea nu va permite funcționarea.
Lampa de polimerizare VALO fără fir se va opri. Cândsunt introduse baterii noi și dacă nivelul de putere
<5% U
<5% U
este restabilit, lampa de polimerizare se va reporni(>95% cădere în U pentru 5 s)(>95% cădere în U pentru 5 s)și va reveni la aceeași stare înainte de pierderea
Nota 2: Recuperare automată
alimentării. Lampa de polimerizare se va recupera în
mod automat în cazul pierderii alimentării.
Frecvența de alimentare(50/60 Hz) câmp magnetic

30 A/m30 A/mIEC 61000-4-8

Câmpurile magnetice cu frecvență înaltă ar trebuisă se situeze la niveluri caracteristice unei locații
obișnuite într-un mediu tipic, rezidențial, de îngrijire asănătății la domiciliu, comercial, spital sau militar.

Nota 1: Lampa de polimerizare nu este dotată cu niciun port sau nicio linie de acces I/O accesibilă.
218

219

Ghid și declarația producătorului pentru imunitatea electromagnetică pentru sistemele de susținere nevitale
Lampa de polimerizare este destinată utilizării în mediul electromagnetic specificat mai jos. Clientul sau utilizatorul trebuie să se asigure că acesta se utilizează într-un
astfel de mediu.
Testul de imunitate

Nivelul de testare IEC
60601

Nivelul deconformitate

Transmitere RF

3 Vrms3 VrmsIEC 61000-4-6

de la 150 kHz până la
80 MHzde la 150 kHz până la
80 MHzRadiații RF

IEC 61000-4-3

3 V/mde la 80 MHz până la
2,5 GHz3 V/mde la 80 MHz până la
2,5 GHz

Ghidul mediului electromagnetic
Echipamentele de comunicații RF portabile și mobile nu ar trebui să fie utilizate în
apropierea vreunei componente a lămpii de polimerizare, inclusiv a cablurilor, cel puțin
până la distanța recomandată de separare calculată din ecuația aplicabilă frecvenței
emițătorului.
Distanța de separare recomandată:de la 80 MHz la 800 MHz

de la 800 MHz la 2,5 GHz
P este puterea de ieșire maximă a emițătorului în wați (W) conform producătoruluiemițătorului și d este distanța de separare recomandată în metri (m).
Câmpurile rezistente de la emițătoarele RF fixe, determinate printr-o verificare
electromagnetică a amplasamentuluia ar trebui să fie mai mici decât nivelul de conformitateîn fiecare interval de frecvențăb.
Interferența poate să apară în vecinătatea echipamentelor marcate cu următorul simbol:

Ghidul și declarația producătorului pentru distanțele recomandate de separare între echipamentele de comunicații RF portabile și mobile și lampa de polimerizare
VALO fără fir
Lampa de polimerizare este destinată utilizării într-un mediu electromagnetic în care perturbațiile radiațiilor RF sunt controlate. Utilizatorul lămpii de polimerizare
poate ajuta la prevenirea interferențelor electromagnetice prin menținerea unei distanțe minime între echipamentul de comunicații RF portabil și mobil (emițătoare)și lampa de polimerizare, așa cum se recomandă mai jos, în funcție de puterea maximă de ieșire a echipamentului de comunicații.
Distanța de separare în funcție de frecvența emițătorului(m)
Putere maximă de ieșire nominală
a emițătorului(P în wați)150 kHz – 80 MHz

80 MHz – 800 MHz

800 MHz – 2.5 GHz

0,01

0,12 metri

0,035 metri

0,07 metri

0,1

0,37 metri

0,11 metri

0,22 metri

1

1,7 metri

0,35 metri

0,7 metri

10

3,7 metri

1,11 metri

2,22 metri

100

11,7 metri

3,5 metri

7,0 metri

Lampa de polimerizare a fost testată în conformitate cu IEC 60601-1-2:2014 și a trecut sub intensități ale câmpului de radiații de 10 V/m între 80 MHz și 2,5 GHz.
Valoarea de 3 Vrms corespunde V1, iar valoarea 10 V/m corespunde cu E1 în formulele de mai sus.
Pentru emițătoarele cu o putere maximă de ieșire care nu figurează mai sus, se poate estima distanța recomandată de separare d în metri (m) folosind ecuația
aplicabilă frecvenței emițătorului, unde P este puterea maximă de ieșire a emițătorului în wați W) conform producătorului emițătorului.
NOTA 1 La 80 MHz și 800 MHz se aplică distanța de separare pentru intervalul de frecvență mai mare.
NOTA 2 Aceste instrucțiuni nu se aplică în toate situațiile. Propagarea electromagnetică este afectată de absorbția și reflexia din structuri, obiecte și persoane.

NOTA 1 La 80 MHz și 800 MHz se aplică intervalul de frecvență mai mare.
NOTA 2 Aceste instrucțiuni nu se aplică în toate situațiile. Propagarea electromagnetică este afectată de absorbția și reflexia din structuri, obiecte și persoane.
Acțiunile unui câmp din emițătoare fixe, cum ar fi stațiile de bază pentru telefoanele radio (mobile/fără fir) și radiourile mobile terestre, radioamatori, difuzare
radio AM și FM și transmisia TV, nu pot fi estimate teoretic cu exactitate. Pentru a evalua mediul electromagnetic datorită emițătoarelor RF fixe, ar trebui să se aibă învedere o verificare a amplasamentului electromagnetic. Dacă intensitatea acțiunii câmpului măsurată în locul în care este folosită lampa de polimerizare depășește
nivelul de conformitate RF corespunzător de mai sus, trebuie respectată lumina de polimerizare VALO fără fir pentru a verifica funcționarea normală. Dacă se observă
performanțe anormale, pot fi necesare măsuri suplimentare, cum ar fi reorientarea sau relocarea lămpii de polimerizare.
bÎn intervalul de frecvență de la 150 kHz la 80 MHz, acțiunea câmpului trebuie să fie mai mică de 3 V/m.
a

220

221

VALO™ GRANDSKVYTVRDZOVACIE SVETLO
Popis produktuVďaka svojmu širokopásmovému spektru je polymerizačná lampa VALO Grand navrhnutá tak, aby polymerizovala všetky produkty vytvrdzované svetlom v rozsahu vlnových dĺžok 385-515
nm podľa ISO 10650. Polymerizačná lampa VALO Grand využíva Ultradent VALO nabíjateľné batérie a nabíjačku batérií. Polymerizačná lampa je navrhnutá tak, aby bola uložená vštandardnom držiaku stomatologickej súpravy alebo sa dá voliteľne namontovať pomocou držiaka, ktorý je súčasťou balenia.

Komponenty produktu VALO Veľké:
•
•
•
•
•
•

1) Pred použitím polymerizačnej lampy nabite batérie (pozrite si časť Údržba batérie).
2) Pred každým použitím umiestnite na polymerizačnú lampu nový ochranný návlek, čím minimalizujete ryhy na šošovke, aby ste dosiahli čo najlepšie výsledky.
• Aby ste zabránili krížovej kontaminácii a zabránili priľnutiu dentálneho kompozitného materiálu k povrchu šošovky a telu svetlovodu, na polymerizačnú lampu VALO sa musí prikaždom použití použiť ochranný návlek schválený spoločnosťou Ultradent. Ochranné návleky sú určené na jednorázové použitie pre pacienta.
Bezdrôtové tienidlo VALO má oválny tvar, pre maximálne využitie je možné ho otáčať a možno ho použiť s priehľadným sterilným sáčkom.

Použitie
Tlačidlá a kontrolky

LegendaTlačidlo(á) napájania(Hore alebo dole)
Kontrolky časovania
Vybitá batéria – Indikátor
Kontrolka režimu/stavu
Tlačidlo zmeny času/režimu
(Timing/Mode Change)(Hore)

Pred použitím, si pre všetky opísané produkty, pozorne prečítajte a porozumejte všetkým pokynom a KBÚ (SDS) informáciám.

Indikácie pre použitie/zamýšľaný účelZdroj osvetlenia pre vytvrdzovanie foto aktivovaných dentálnych obnovujúcich materiálov a lepidiel.

Upozornenia a opatrenia
2. Riziková skupina
UPOZORNENIE UV žiarenie je vyžarované z tohto výrobku. Môže dôjsť k podráždeniu očí alebo kože. Použite vhodné tienenie.
UPOZORNENIE Možné nebezpečné optické žiarenie emitované z tohto výrobku. Nepozerajte sa na zapnuté svetlo. Môže byť škodlivé pre oči.
• NEPOZERAJTE sa priamo do výstupu lampy. Pacient, lekár a asistenti by mali pri použití polymerizačnej lampy vždy používať UV ochranu očí oranžovej farby.
• Aby sa zabránilo riziku úrazu elektrickým prúdom, nie sú povolené žiadne úpravy tohto zariadenia. Používajte iba priložený napájací zdroj Ultradent VALO a zástrčkové adaptéry. Ak sú
tieto komponenty poškodené, nepoužívajte ich, zavolajte zákaznícky servis Ultradent a objednajte si výmenu.
• Prenosné vysokofrekvenčné komunikačné zariadenie môže znížiť výkon, ak sa použije vo vzdialenosti menšej ako 30 cm (12 palcov).
• Používajte iba autorizované príslušenstvo, káble a napájacie zdroje, aby ste predišli nesprávnemu fungovaniu, zvýšeným elektromagnetickým emisiám alebo zníženiu
elektromagnetickej odolnosti (pozrite si časť Elektromagnetické emisie).
• Aby ste predišli riziku vzniku elektrického požiaru spojeného s manipuláciou s batériami:
⸰ NEMIEŠAJTE nabíjateľné batérie s nenabíjateľnými batériami alebo inými typmi batérií.
⸰ NEPOKÚŠAJTE sa nabíjať nenabíjateľné batérie. Nabíjanie nenabíjateľných batérií môže viesť k zraneniu alebo poškodeniu majetku.
⸰ Batérie, kontakty batérií, nabíjačku ani sieťový adaptér NEAUTOKLÁVUJTE ani nestriekajte tekutinou akéhokoľvek druhu. Ak sa na kontaktoch nabíjačky batérie objaví korózia,zavolajte na zákaznícky servis Ultradent a objednajte si výmenu.
⸰ NENABÍJAJTE batérie v blízkosti horľavých materiálov.
⸰ NENECHÁVAJTE nabíjačku v klinickej prevádzke.
• Aby ste predišli riziku zranenia, NEPOUŽÍVAJTE batérie, ktoré sú skorodované (hrdzavé), preliačené, zapáchajú alebo vytekajú, majú roztrhnutý alebo chýbajúci obal alebo sú inak
poškodené. Zavolajte na zákaznícky servis Ultradent a objednajte si náhradné batérie.
• Používajte len odporúčané batérie. Alternatívne batérie môžu spôsobiť nesprávnu funkciu.
• Aby ste predišli riziku tepelného podráždenia alebo poranenia, vyhnite sa nepretržitým následným cyklom polymerizovania a nevystavujte mäkké tkanivá ústnej dutiny v tesnej blízkosti
na viac ako 10 sekúnd v žiadnom režime. Ak sú potrebné dlhšie časy vytvrdzovania, použite viacero cyklov vytvrdzovania s prestávkami medzi cyklami alebo použite produkt s dvojitým
vytvrdzovaním, aby ste predišli zahrievaniu mäkkého tkaniva.
• Buďte opatrní pri liečbe pacientov, ktorí trpia nežiaducimi fotobiologickými reakciami alebo citlivosťou, pacientov, ktorí podstupujú chemoterapiu alebo pacientov liečených
fotosenzibilizujúcimi liekmi.
• Toto zariadenie môže byť citlivé na silné magnetické alebo statické elektrické polia, ktoré by mohli narušiť jeho naprogramovanie. Ak máte podozrenie, že k tomu došlo, na chvíľu
odpojte jednotku a potom ju znova zapojte do zásuvky.
• NEUTIERAJTE polymerizačnú lampu žieravinami alebo abrazívnymi čistiacimi prostriedkami, nevkladajte ju do autoklávu ani ju neponárajte do akéhokoľvek druhu ultrazvukovéhokúpeľa, dezinfekčného prostriedku, čistiaceho roztoku alebo kvapaliny. Nedodržanie priložených pokynov na údržbu môže spôsobiť nefunkčnosť polymerizačného svetla.
222

Postupné pokyny
Príprava

Bezdrôtové tienidlo VALO:

Fotografia(Dole)

Aby ste predišli poškodeniu zariadenia, NEVKLADAJTE prsty, nástroje ani iné predmety do priestoru pre batérie polymerizačnej lampy.
Aby ste predišli poškodeniu zariadenia, NEPOKÚŠAJTE sa čistiť zlaté kontakty ani žiadnu časť priestoru pre batérie. Ak máte obavy, zavolajte zákaznícky servis Ultradent.
Aby sa predišlo riziku krížovej kontaminácie, ochranné návleky sú určené na jednorázové použitie pre pacienta.
Aby ste znížili riziko korózie, po použití odstráňte ochranný návlek.
Na zníženie rizika nedostatočne vytvrdených živíc nepoužívajte polymerizačnú lampu s poškodenou šošovkou.

Poznámka:

1 – Polymerizačná lampa VALO Grand4 – Ultradent VALO nabíjateľné batérie
1 – Nabíjačka batérií Ultradent VALO s 12 VDC napájacím zdrojom pre lekárske účely
1 – Vzorkové balenie ochranných návlekov VALO Barrier Sleeve
1 – Svetelný štít VALO Grand
1 – Držiak na povrchovú montáž polymerizačnej lampy s obojstrannou lepiacou páskou

Prehľad ovládacích prvkov:

•
•
•
•
•

1) Každý režim napájania sa používa na vytvrdzovanie zubných materiálov s fotoiniciátormi. Odporúčané časy vytvrdzovania nájdete v príručke Stručnej príručke režimov.
POZNÁMKA: Vytvrdzovacie svetlo je naprogramované tak, aby sa postupne prepínalo z štandardného výkonu na vysoký výkon plus do režimu extra výkonu. Ak chcete napríklad prepnúť zrežimu štandardného výkonu do režimu extra výkonu, je potrebné prejsť do režimu vysokého výkonu plus potom do režimu extra výkonu.
2) Vytvrdzovacie svetlo uloží naposledy použitý časový interval a režim, ktorý sa použije pri každej zmene režimov alebo pri vybratí batérií.

ObsluhaREŽIM VYTVRDZOVANIA: Štandardný režim napájaniaČASOVÉ INTERVALY: 5, 10, 15, a 20 sekúnd.
• Vytvrdzovacie svetlo je predvolené pre tento režim, keď je PÔVODNE zapnuté. Kontrolka režimu/stavu bude svietiť nazeleno a rozsvietia sa štyri zelené časové kontrolky, ktoré
označujú režim štandardného výkonu.
• Ak chcete zmeniť časové intervaly, rýchlo stlačte tlačidlo čas/režim (Time/Mode).
• Pre vytvrdzovanie stlačte tlačidlo napájania (Power). Ak chcete ukončiť vytvrdzovanie pred dokončením časového intervalu, znova stlačte tlačidlo napájania (Power).

REŽIM VYTVRDZOVANIA: Režim vysokého výkonu

ČASOVÉ INTERVALY: 1, 2, 3, a 4 sekúnd.
• V režime štandardného výkonu stlačte a podržte tlačidlo zmeny času/režimu (Time/Mode Change) na 2 sekundy a pustite ho. Kontrolka režimu/stavu bude svietiť na oranžovo arozsvietia sa a budú blikať štyri zelené časové kontrolky, indikujúce režim vysokého výkonu.
• Ak chcete zmeniť časové intervaly, rýchlo stlačte tlačidlo čas/režim (Time/Mode).
• Pre vytvrdzovanie stlačte tlačidlo napájania (Power). Ak chcete ukončiť vytvrdzovanie pred dokončením časového intervalu, znova stlačte tlačidlo napájania (Power).
• Pre návrat do režimu štandardného výkonu, stlačte a podržte tlačidlo zmeny času/režimu (Time/Mode Change) na 2 sekundy a pustite, toto sa prepne do režimu extra výkonu. Opäťho stlačte a podržte na 2 sekundy a pustite. Kontrolka režimu/stavu bude svietiť nazeleno a rozsvietia sa štyri zelené časové kontrolky, ktoré označujú režim štandardného výkonu.

REŽIM VYTVRDZOVANIA: Režim extra výkonu

ČASOVÝ INTERVAL: Len 3 sekundy (Poznámka: Režim extra výkonu má 2-sekundové bezpečnostné oneskorenie na konci každého vytvrdzovacieho cyklu na obmedzenie zahrievania
počas nadväzného vytvrdzovania. Na konci oneskorenia pípnutie signalizuje, že jednotka je pripravená na ďalšie použitie).
• V režime štandardného výkonu stlačte tlačidlo zmeny času/režimu (Time/Mode Change) na 2 sekundy, pustite ho a znova ho podržte na 2 sekundy a pustite. Kontrolka režimu/stavu
bude svietiť na oranžovo a rozsvietia sa a budú blikať tri zelené časové kontrolky, indikujúce režim extra výkonu.
• Pre vytvrdzovanie stlačte tlačidlo napájania (Power). Ak chcete ukončiť vytvrdzovanie pred dokončením časového intervalu, znova stlačte tlačidlo napájania (Power).
• Pre návrat do režimu štandardného výkonu, stlačte a podržte tlačidlo zmeny času/režimu (Time/Mode Change) na 2 sekundy a pustite. Kontrolka režimu/stavu bude svietiť nazelenoa rozsvietia sa zelené časové kontrolky, ktoré označujú režim štandardného výkonu.

Režim spánku: Vytvrdzovacie svetlo sa prepne do režimu spánku po 60 sekundách nečinnosti, čo je indikované pomalým blikaním kontrolky režimu/stavu. Zdvihnutie alebo dotknutie
sa jednotky zobudí vytvrdzovacie svetlo a automaticky ho vráti na posledné použité nastavenie. Pre maximalizáciu životnosti batérie, ponechajte vytvrdzovacie svetlo v pokoji, keď sa
nepoužíva.

Čistenie

1) Po každom pacientovi odstráňte použitý sterilný sáčok do štandardného odpadu.
2) Pozri časť Spracovanie.

Pokyny pre montážnu konzolu

1) Konzola by sa mala namontovaná na rovný, bez olejový povrch.
2) Povrch očistite liehom.
3) Odlepte lepiacu pásku konzoly.
4) Konzolu umiestnite tak, aby sa vytvrdzujúce svetlo pri vyňatí nadvihlo nahor. Pevne zatlačte na miesto.

223

Sprievodca rýchlym režimom:Štandardný výkon
1000 mW/cm2

Režim

Vysoký výkon plus
1600 mW/cm2

Extra výkon
3200 mW/cm2

Tlačidlo napájania (Power)

Ako nabíjať batérie:1) Nabíjačku a batérie umiestnite na miesto, kde sa nedostanú do kontaktu s kvapalinami alebo dezinfekčnými sprejmi.
2) Batérie vložte do nabíjačky tak, aby kladný (+) pól smeroval k indikátorom na nabíjačke.
3) Zelené indikátory znázorňujú, že sú batérie pripravené na použitie.
4) Nabitie batérií bude trvať 1-3 hodiny. Nechajte batérie v nabíjačke, kým nebudú pripravené na použitie.
POZNÁMKA: Ak sa červený indikátor na nabíjačke nerozsvieti na zeleno, keď sa batérie nabíjajú dlhšie ako tri hodiny, batéria môže byť chybná (poškodená) a nemožno ju nabíjať.
Vyskúšajte novú batériu alebo zavolajte na zákaznícky servis Ultradent a objednajte si novú súpravu nabíjateľných batérií.

LED kontrolky režimu/času

Tlačidlá času (Time)Možnosti času

5s

10s

15s

20s

1s

2s

3s

4s

Iba 3sPre zmenu časuRýchlo stlačte a pustite tlačidlo času (Time) pre prepínanie časových možnosti.

Pre zmenu režimu

Stlačte a podržte tlačidlo času (Time) na 2 sekundy a pustite ho. Vytvrdzovacie svetlo sa prepne na nasledujúci režim.

LegendaSvietiace LEDkyRežim vytvrdzovaniaVýkon* (mW)Ožiarenie* (mW/cm2)

Celkový čas expozície (sekundy)

Energia** (Jouly)Štandardný

970

900

5/10/15/20

4,8 J/9,7 J/14,5 J/19,4 J

Vysoký výkon Plus

1620

1500

1/2/3/4

1,6 J/3,2 J/4,8 J/6,5 J

Xtra

2260

2100

3

6,8 J

*Nominálny výkon vo vzdialenosti 2 mm od špičky šošovky po horný povrch kompozitu.
**Hodnoty v sekcii celková energia na cyklus (jouly) sú zaokrúhlené nahor na najbližšiu desatinu.

Varovanie úrovne napájania

Varovanie teploty

Servisné varovanie

LED VarovanieVymeňte batérie

Nechajte vychladnúťPre opravu zavolajte na zákaznícky
servisPre opravu zavolajte na zákaznícky
servis• Žiadny zvuk
• Bliká každé 2 sekundy• Umožňuje prevádzku

• 3 nadväzné pípnutia• Jedno rýchle bliknutie• Zakazuje operácie• Vybitá batéria: pomalé blikanie• Vypnúť: 3 pípnutia, blikanie• Zakazuje prevádzku

• 3 pípnutia• Pomalé blikanie• Zakazuje prevádzku

Ako vymeniť/vložiť batérie:1) Odstráňte zadný kryt otočením proti smeru hodinových ručičiek o štvrtinu otáčky.
2) Vyberte batérie.
3) Vložte nové batérie najskôr kladným (+) pólom.
4) Znova nasaďte zadný kryt tak, že ho zarovnáte a jemne zatlačíte a súčasne otočíte v smere hodinových ručičiek. Po úplnom nasadení kryt zacvakne.
5) Jednotka je pripravená na použitie.
Vybité batérie: polymerizačná lampa používateľovi signalizuje, že je čas vymeniť batérie, keď indikátor vybitej batérie bliká na červeno. Ak je nabitie batérie príliš nízke, zaznie varovný zvuks tromi pípnutiami a polymerizačná lampa neumožní ďalšiu prevádzku, kým sa batérie nenabijú alebo sa nevložia nové batérie. (Pozrite si Rýchleho sprievodcu varovaním)

Blikajúce LEDky

Sprievodca rýchlym vytvrdzovaním:

Sprievodca rýchlym varovaním:

prilepený kompozit opatrne odstránili. Nepoužívajte nástroje, ktoré môžu poškodiť šošovku.
2) Svetlomery sa značne líšia a sú navrhnuté pre špecifické svetlovodné hroty a šošovky. Ultradent odporúča pravidelne kontrolovať výstup v režime Štandardný výkon. POZNÁMKA:Skutočný číselný výstup bude skreslený v dôsledku nepresnosti bežných svetlomerov a vlastného LED púzdra v polymerizačnej lampeÚdržba batérie
Nabíjanie a výmena batérií
Polymerizačná lampa sa dodáva so 4 nabíjateľnými lítium-železo-fosfátovými batériami. Ultradent odporúča používať iba VALO nabíjateľné batérie.

Čas nabíjania a výdrž batérie: Životnosť úplne nabitej batérie v polymerizačnej lampe závisí od nastavenia Mode/Time (Režimu/Časového intervalu), typu batérie, frekvencie používaniaa účinnosti LED.
• Odporúčaný interval nabíjania: dobíjajte podľa potreby alebo keď sa rozsvieti indikátor vybitej batérie. Priemerná dĺžka života závisí od používania.
• Extra batérie: Polymerizačná lampa sa dodáva so 4 nabíjateľnými batériami.
Opravy výrobcu
1) Opravy smie vykonávať iba autorizovaný servisný personál. Ultradent poskytne servisnému personálu dokumentáciu na vykonanie opráv.
2) Pri odosielaní kompletného balenia na opravu, servis alebo kalibráciu batérie vždy vyberte z polymerizačnej lampy a nabíjačky. Batérie, nabíjačku, adaptér a polymerizačnú lampuzabaľte oddelene do vratnej krabice.
3) Batérie posielajte v súlade s miestnymi predpismi.
ZárukaUltradent Products, Inc. („Ultradent“) zaručuje, že tento produkt bude po dobu 5 rokov od dátumu zakúpenia fungovať podľa prevádzkových pokynov priložených k produktu, (i)zodpovedá vo všetkých materiálnych ohľadoch špecifikáciám uvedeným v dokumentácii Ultradent priloženej k produktu; a (ii) neobsahuje chyby materiálu a spracovania.
Táto obmedzená záruka je neprenosná a vzťahuje sa výlučne na pôvodného kupujúceho a nevzťahuje sa na následných vlastníkov produktu. Táto obmedzená záruka sa nevzťahuje
na žiadne ďalšie súčasti príslušenstva, ako sú, ale nie výlučne, batérie, nabíjačky, adaptéry alebo adaptívne šošovky. Táto obmedzená záruka je neplatná, ak produkt zlyhá alebo jepoškodený v dôsledku nedbanlivosti, zneužitia, nesprávneho použitia, nehody, úpravy, zásahu, zmeny alebo nedodržania príslušných pokynov na použitie. Táto záruka sa nevzťahuje
napríklad na produkt, ktorý spadne a poškodí sa. Na uplatnenie tejto obmedzenej záruky je potrebné spolu s chybným produktom predložiť spoločnosti Ultradent doklad o kúpe (napr.
účtenku o predaji alebo podobnú dokumentáciu).
Chybný produkt, ktorý spĺňa záručné podmienky uvedené v tomto dokumente, bude podľa vlastného uváženia spoločnosti Ultradent buď opravený alebo vymenený. Zodpovednosť
Ultradent za produkt v žiadnom prípade nepresiahne kúpnu cenu zaplatenú kupujúcim. Ultradent za žiadnych okolností nezodpovedá za žiadne nepriame, náhodné, predvídané,
nepredvídané, špeciálne alebo následné škody vyplývajúce z používania tohto produktu alebo v súvislosti s ním.

Spracovanie
Tento výrobok je určený na použitie s ochranným návlekom. Údržbu a všeobecné čistenie nájdete vyššie. Skladovanie a likvidácia

Ak vytvrdzovacie svetlo skladujete po obdobie dlhšie ako 2 týždne, alebo ho balíte na cestu, vždy vyberte batérie. Ak zostanú batérie v prístroji dlhší čas bez nabíjania, môžu sa staťnefunkčnými alebo nenabíjateľnými. Batérie neskladujte pri teplotách nad 40°C (104°F) alebo na priamom slnečnom svetle.

ÚdržbaVšeobecné čistenie polymerizačnej lampy
Po každom použití navlhčite gázu alebo mäkkú handričku schváleným prostriedkom na dezinfekciu povrchov a utrite povrch a šošovku. Neautorizované čistiace prostriedky môžuspôsobiť poškodenie polymerizačnej lampy.
PRIJATEĽNÉ ČISTIACE PROSTRIEDKY:• 70 % izopropylalkohol• 70 % etanol

Všeobecné čistenie svetelného štítu:
Svetelný štít VALO Cordless vyčistite pomocou akéhokoľvek dezinfekčného prostriedku na povrchy. NEAUTOKLÁVUJTE.
OpravaÚdržba vykonávaná používateľom
1) Aby ste zabránili priľnutiu zubného kompozitu k povrchu šošovky, použite ochranný návlek. Ak je to potrebné, použite dentálny nástroj z plastu alebo nehrdzavejúcej ocele, aby ste
224

Skladovanie a preprava vytvrdzovacieho svetla:• Teplota: +10°C do +40°C (+50°F do +104°F)• Relatívna vlhkosť: 10% až 95%• Okolitý tlak: 500 hPa až 1060 hPa
Pri likvidácii elektronického odpadu (t. J. vytvrdzovacích svetiel, nabíjačiek, batérií a napájacích zdrojov) dodržiavajte miestne smernice o odpade a recyklácii.

Technické pokyny
Príslušenstvo
Položka

Informácie o CE

Bezdrôtové tienidlo VALO

225

Položka

Informácie o CE

Technické informácie/údaje

Účinné vlnové dĺžky kompozitného vytvrdzovania:

Nabíjačka VALOSPEKTRÁLNY ŽIAROVÝ VÝKON (mW/nm)

Batérie VALOEFEKTÍVNE KOMPOZITNÉ VYTVRDZUJÚCE PÁSY VLNOVEJ DĹŽKY
(Využiteľný rozsah vlnovej dĺžky: 385-515 nm)Účinná nižšia
vlnové pásmo
395-415 nm

Efektívne pásmohorných vlnových
dĺžok
440-480 nm

Xtra
Vysoký výkon
ŠtandardnýWAVEL ENGTH (nm)Atribút

Informácie/špecifikácieŠošovkaPriemer 11,7 mm

Rozsah vlnových
dĺžok• Použiteľný rozsah vlnových dĺžok: 385 - 515 nm• Maximálne vlnové dĺžky: 395 - 415 nm a 440 - 480 nm
Porovnávacia tabuľka nominálnej intenzity žiarenia
Merací prístrojTabuľka intenzity
svetla226‡ Analyzátor Gigahertzovho spektraŽiarenie

Celkový výkon

Clona merača

15 mm

15 mm

Štandardný výkon
(±10%)

900 mW/cm²

970 mW

Vysoký výkon plus(±10%)

1500 mW/cm²

1615 mW

Extra výkon (±10%)2100 mW/cm²

Vyžarovanie sa bude líšiť v závislosti odschopnosti prístroja, metódy merania aumiestnenia lampy.
‡ Vyžarovanie je v súlade s normou ISO 10650
pri meraní pomocou spektrálneho analyzátora
Gigahertz-Optik.

2260 mW

Vytvrdzova-cie
svetlo VALO Veľké

Hodnotenie: IEC 60601-1 (Bezpečnosť), IEC
60601-1-2 (EMC)

Hmotnosť:• S batériami: 6 uncí (190 gramov)• Bez batérií: 5 uncí (150 gramov)Rozmery: (8 x 1,28 x 1,06) palcov, (203 x 32,5 x 27) mm

Napájanie nabíjačkyVýkon - 12VDC pri 500mA
Príkon - 100VAC do 240VAC
Ultradent P/N 5930 Napájacia nabíjačka VALO suniverzálnymi zástrčkami

Hodnotenie: IEC 60601-1 (Bezpečnosť)Dĺžka kábla - 6 stôp (1,8 metra)
Napájacia nabíjačka VALO je napájacím zdrojom na lekárske účely II. triedy azabezpečuje izoláciu od hlavného napájania

Nabíjačka VALOInteligentná nabíjačka batérií VALO 3.6VDC fosforečnan lítno-železitý:• Automatické vypnutie po úplnom nabití• Automatická detekcia chybných batérií• Ochrana: Tepelná, prebitie, skrat, obrátená polarita⸰ Červená LED - Nabíja sa
⸰ Zelená LED - Prázdna alebo plne nabitá
⸰ LED nesvieti - Skrat• Doba nabíjania: 1 - 3 hodiny
Hodnotenie: CE, OEEZ
227

Rôzne informácie

Atribút

Informácie/špecifikácie

Batérie VALO

Nabíjateľné: Bezpečné, chemické, lítium-železo-fosfátové (LiFePO4) RCR123A•
Pracovné napätie: 3,2 VDC, 400 mAh (potrebná watthodina)
Normy: UL 62133-2, CE, RoHS, WEEEČlánky sú dimenzované na >2000 cyklov nabitia/vybitia pred dosiahnutím 70 % svojej kapacity. Batérie je odporúčané vymieňať
každých 3-5 rokov, avšak ich životnosť sa môže líšiť v závislosti od ich používania. V prípade potreby ich vymeňte.

Prevádzkové
podmienky

Teplota: +10°C do +32°C (+50°F do +90°F)
Relatívna vlhkosť: 10% až 95%
Okolitý tlak: 700 hPa až 1060 hPa

Pracovný cyklus

Vytvrdzovacie svetlo je určené na krátkodobú prevádzku. Pri maximálnej okolitej teplote (32°C) 1 minútu ZAPNUTÉ (ON) pri
nadväznej cyklácií, 30 minút VYPNUTÉ (OFF) (doba chladenia).

Riešenie problémov
Ak vyššie navrhnuté riešenia problém nevyriešia, zavolajte Ultradent na čísle 800 552 5512. Mimo Spojených štátov kontaktujte svojho distribútora Ultradent alebo
dentálneho predajcu.
Problém

Smernice a vyhlásenie výrobcu pre elektromagnetické emisie
Vytvrdzovacie svetlo je určené na použitie v elektromagnetickom prostredí špecifikovanom nižšie. Zákazník alebo užívateľ by sa mali uistiť, že sa používa v takomto
prostredí.
Emisný testDodržiavanie

Elektromagnetické prostredie - smernica

RF emisie
CISPR 11

1. skupina

Vytvrdzovacie svetlo využíva batérie a nie je ovplyvňované EMI, RF alebo potlačením prepätia.

RF emisie
CISPR 11

Trieda B

Vytvrdzovacie svetlo využíva elektrickú a elektromagnetickú energiu len pre svoje vnútorné funkcie.
Preto sú akékoľvek RF emisie veľmi nízke a pravdepodobne nespôsobujú rušenie v blízkych
elektronických zariadeniach.

Harmonické emisie
IEC 61000-3-2Nevzťahuje sa

Testy harmonických emisií a kolísania napätia sa na vytvrdzovacie svetlo nevzťahujú, pretože jenapájané z batérie.

Kolísanie napätia/blikanie emisií Nevzťahuje sa
IEC 61000-3-3Možné riešenia

Svetlo sa nezapne

1) Stlačte tlačidlo zmeny času/režimu (Time/Mode Change) alebo tlačidlo napájania (Power) pre prebudenie z úspornéhorežimu.
2) Skontrolujte červený indikátor stavu vybitia batérie.
3) Skontrolujte, či sú nové batérie správne vložené do zariadenia.
4) Ak blikajú červené a žlté výstražné LED indikátory, znamená to, že vytvrdzovacie svetlo dosiahlo vnútornú hranicu
bezpečnostnej teploty. Vytvrdzovacie svetlo nechajte vychladnúť na 10 minút alebo použite chladný vlhký uterák na rýchle
ochladenie zariadenia.
5) Ak červený LED indikátor bliká a nepretržite pípa, zavolajte na zákaznícky servis Ultradent pre opravu.

Svetlo nezostáva zapnuté popožadovanú dobu

1) Skontrolujte režim a časové kontrolky pre správne nastavenie času.
2) Skontrolujte indikátor stavu vybitia batérie.
3) Skontrolujte, či sú nové batérie správne vložené do zariadenia.

Svetlo nevytvrdzuje živice správne

1) Skontrolujte šošovky pre zvyškovú vytvrdenú živicu/kompozity.
2) Použite vhodnú oranžovú ochranu očí pred UV žiarením a overte, či LED kontrolky fungujú.
3) Skontrolujte úroveň výkonu pomocou expozimetra. Ak používate expozimeter, Ultradent odporúča kontrolu v režime
štandardného výkonu.
POZNÁMKA: Skutočný číselný výstup bude skreslený v dôsledku nepresnosti bežných expozimetrov a vlastného balenia LEDpoužívaného vytvrdzovacím svetlom. Expozimetre sa veľmi líšia a sú určené pre špecifické svetlovodné špičky a šošovky.
4) Skontrolujte dátum spotreby vytvrdzovacej živice.
5) Uistite sa, že je dodržaná správna technika podľa odporúčaní výrobcu.

Batérie sa nenabijú

1) Uistite sa, že batérie sú vložené v nabíjačke v správnej polohe a nechajte batérie nabíjať 1-3 hodiny.
2) Ak sa červená kontrolka na nabíjačke nezmení na zelenú, zavolajte zákaznícky servis Ultradent, aby ste si objednalináhradné batérie a/alebo nabíjačku.
3) Ak nie je na nabíjačke viditeľná ani zelená ani červená kontrolka, zavolajte zákaznícky servis Ultradent a objednajte si
alebo vymeňte nabíjačku a/alebo sieťový adaptér.

Nabíjačka nenabíja batérie

1) Skontrolujte, či je nabíjačka zapojená a či je sieťový adaptér pripojený do fungujúcej elektrickej zásuvky.
2) Ak nie je na nabíjačke viditeľná ani zelená ani červená kontrolka, zavolajte zákaznícky servis Ultradent a objednajte si novú
nabíjačku a/alebo sieťový adaptér.

Nemožno zmeniť režim alebo časové
intervaly

1) Podržte stlačené obe tlačidlá čas/režim (Time/Mode) a napájanie (Power), až kým séria pípnutí neindikuje, že sa
vytvrdzovacie svetlo odomklo.

Vytvrdzovacie svetlo je vhodné na použitie vo všetkých prevádzkach, vrátane domácich zariadenía zariadení priamo pripojených na verejnú nízkonapäťovú sieť, ktorá zásobuje budovy na domácepoužitie.

Smernice a vyhlásenie výrobcu pre elektromagnetickú imunitu
Vytvrdzovacie svetlo je určené na použitie v elektromagnetickom prostredí špecifikovanom nižšie. Zákazník alebo užívateľ by sa mali uistiť, že sa používa v takomto
prostredí.
Test IMUNITY

Testovacia úroveň IEC 60601

Úroveň súladu

Pokyny pre elektromagnetické prostredie

Elektrostatický výboj (ESD)± 8 kV kontakt± 15 kV vzduch± 8 kV kontakt± 15 kV vzduch

Fyzické prostredie by malo byť obmedzené na
nasledovné:
1) Kód IP: IP20
2) Neponárajte do kvapaliny.
3) Nepoužívajte v blízkosti horľavých plynov.
Jednotka nie je APG a AP.
4) Rozsah vlhkosti pri skladovaní: 10% - 95%
5) Rozsah skladovacích teplôt: 10°C - 40°C

IEC 61000-4-2

Elektrický rýchly prechod/impulz
IEC 61000-4-4± 2 kV pre vedenia napájania± 2 kV pre vedenia napájania± 1 kV pre vedenia príkonu/výkonu 1. Poznámka: Vytvrdzovacie svetlo
nemá žiadne I/0 porty

Prepätie
IEC 61000-4-5± 1 kV vedenie k vedeniu± 2 kV vedenia k zemi

± 1 kV vedenie k vedeniu± 2 kV vedenia k zemi

Napätie, poklesy, skraty,
prerušenia a variácie vstupných
vedení napájania<5% U(> 95% pokles v U po dobu
0,5 cyklu)<5% U(> 95% pokles v U po dobu
0,5 cyklu)

IEC 61000-4-11

40% U(60% pokles v U po dobu 5 cyklov)

40% U(60% pokles v U po dobu 5 cyklov)

70% U(30% pokles v U po dobu 25
cyklov)

70% U(30% pokles v U po dobu 25
cyklov)<5% U(> 95% pokles v U po dobu 5 s)

<5% U(> 95% pokles v U po dobu 5 s)
2. Poznámka: Vlastné obnovuje

30 A/m30 A/mFrekvencia napájania(50/60 Hz) magnetické pole
IEC 61000-4-8

Vytvrdzovacie svetlo je napájané z batérie a nie jemožné ho pripojiť sa na sieť striedavého napätia.
Pretože vytvrdzovacie svetlo je napájané z batérie,nie je vystavené elektrickému rušeniu, prepätiam,poklesom napätia, skratom, prerušeniam alebozmenám napájania siete striedavého napätia.
Súčasť príslušenstva, nabíjačka batérií, môže podliehať
vyššie uvedenému, ale je oddelená od a nie jekritická pre prevádzku bezdrôtového vytvrdzovacieho
svetla VALO.
Ak napätie batérie vytvrdzovacieho svetla klesne na
4VDC, zariadenie neumožní prevádzku. Bezdrôtové
vytvrdzovacie svetlo VALO sa vypne. Po vložení nových
batérií a opätovnom obnovení správnej úrovnevýkonu, vytvrdzovacie svetlo sa reštartuje a vráti sa
do rovnakého stavu ako pred stratou napätia. V
prípade straty napájania sa vytvrdzovacie svetlo
samočinne obnoví.
Magnetické polia frekvencie napájania by mali byť naúrovniach charakteristických pre typické umiestneniev typickom obytnom, domácom zdravotníckom,
komerčnom, nemocničnom alebo vojenskom
prostredí.

1. Poznámka: Vytvrdzovacie svetlo nie je vybavené žiadnymi portami ani žiadnymi prístupovým I/O vedením.

228

229

1. Poznámka: Vytvrdzovacie svetlo nie je vybavené žiadnymi portami ani žiadnymi prístupovým I/O vedením.
Vytvrdzovacie svetlo je určené na použitie v elektromagnetickom prostredí špecifikovanom nižšie. Zákazník alebo užívateľ by sa mali uistiť, že sa používa v takomto
prostredí.
Test IMUNITY

Testovacia úroveň IEC
60601

Úroveň súladu

Vedenie RF

3 Vrms3 VrmsIEC 61000-4-6

150 kHz do 80 MHz

150 kHz do 80 MHz

Pokyny pre elektromagnetické prostredie
Prenosné a mobilné RF komunikačné zariadenia by sa nemali používať bližšie k žiadnej časti
vytvrdzovacieho svetla, vrátane káblov, než je odporúčaná vzdialenosť, vypočítaná z rovnicevzťahujúcej sa na frekvenciu vysielača.
Odporúčaná vzdialenosť:80 MHz do 800 MHz

800 MHz do 2,5 GHzVyžarovaná RF

3 V/m

3 V/m

P je maximálny výkonový výkon vysielača vo wattoch (W) podľa výrobcu vysielača a d jeodporúčaná vzdialenosť v metroch (m).
Intenzita poľa z pevných RF vysielačov, ako je určené elektromagnetickým prieskumom na
miestea, by mala byť nižšia ako úroveň zhody v každom frekvenčnom rozsahub.

IEC 61000-4-3

80 MHz do 2,5 GHz

80 MHz do 2,5 GHzInterferencie sa môžu vyskytnúť v blízkosti zariadení označených nasledujúcim symbolom:

Smernice a vyhlásenie výrobcu pre odporúčané vzdialenosti medzi prenosnými a mobilnými RF komunikačnými zariadeniami a bezdrôtovým vytvrdzovacím svetlom
VALO
Vytvrdzovacie svetlo je určené na použitie v elektromagnetickom prostredí, v ktorom sa riadi vyžarované rušenie RF. Používateľ vytvrdzovacieho svetla môže pomôcť
zabrániť elektromagnetickému rušeniu udržiavaním minimálnej vzdialenosti medzi prenosnými a mobilnými RF komunikačnými zariadeniami (vysielačmi) a
vytvrdzovacím svetlom ako sa odporúča nižšie podľa maximálneho výstupného výkonu komunikačného zariadenia.
Odstupová vzdialenosť podľa frekvencie vysielača(v metroch)
Menovitý maximálny výstupný výkon
vysielača(P vo wattoch)

150 kHz – 80 MHz

80 MHz – 800 MHz

800 MHz – 2,5 GHz

0,01

0,12 metra

0,035 metra

0,07 metra

0,1

0,37 metra

0,11 metra

0,22 metra

1

1,7 metra

0,35 metra

0,7 metra

10

3,7 metra

1,11 metra

2,22 metra

100

11,7 metra

3,5 metra

7,0 metrov

Vytvrdzovacie svetlo bolo testované podľa normy IEC 60601-1-2:2014 a prešlo v rámci intenzity vyžarovaného poľa 10 V/m medzi 80 MHz a 2,5 GHz. Hodnota 3Vrms
zodpovedá hodnote V1 a hodnota 10 V/m zodpovedá E1 vo vyššie uvedených vzorcoch.
V prípade vysielačov s maximálnym výstupným výkonom, ktoré nie sú uvedené vyššie, možno odporúčanú vzdialenosť d v metroch (m) odhadnúť pomocou rovnicepoužitej pre frekvenciu vysielača, kde P je maximálny výkon vysielača vo wattoch (W) podľa výrobcu vysielača.
1. POZNÁMKA Pri frekvenciách 80 MHz a 800 MHz platí vzdialenosť pre vyššie frekvenčné pásmo.
2. POZNÁMKA Tieto smernice nemusia platiť vo všetkých situáciách. Elektromagnetické šírenie je ovplyvnené absorpciou a odrazom od štruktúr, objektov a ľudí.

1. POZNÁMKA Pri 80 MHz a 800 MHz platí vyšší frekvenčný rozsah.
2. POZNÁMKA Tieto smernice nemusia platiť vo všetkých situáciách. Elektromagnetické šírenie je ovplyvnené absorpciou a odrazom od štruktúr, objektov a ľudí.
aIntenzity poľa z pevných vysielačov, ako sú základňové stanice pre rádio (mobilné/bezdrôtové) telefóny a pozemné mobilné rádia, amatérske rádio, rádiové vysielaniev pásme AM a FM a televízne vysielanie, nemožno predpovedať s teoretickou presnosťou. Na posúdenie elektromagnetického prostredia v dôsledku pevných RF
vysielačov je potrebné zvážiť elektromagnetický prieskum miesta. Ak nameraná intenzita poľa v mieste, kde sa používa vytvrdzovacie svetlo, prevyšuje príslušnú úroveň
zhody RF, bezdrôtové vytvrdzovacie svetlo VALO musí byť sledované, aby sa overila normálna prevádzka. Ak sa zistí abnormálny výkon, môžu byť potrebné ďalšie
opatrenia, ako napríklad zmena orientácie alebo premiestnenie vytvrdzovacieho svetla.
b
Vo frekvenčnom pásme 150 kHz až 80 MHz by intenzity poľa mali byť menšie ako 3 V/m.

230

231

VALO™ GRAND

SL

polimerizacijska lučka
Opis izdelka

S svojim širokopasovnim spektrom je polimerizacijska lučka VALO Grand zasnovana za polimerizacijo vseh svetlobno polimeriziranih izdelkov v območju valovnih dolžin 385–515 nm po
ISO 10650. Polimerizacijska lučka VALO Grand uporablja polnilne baterije Ultradent VALO in polnilnik baterij. Polimerizacijska lučka je zasnovana tako, da leži v standardnem nosilcu zobne
enote ali pa jo je mogoče namestiti po meri z uporabo nosilca, ki je priložen kompletu.

Komponente brezžične lučke Pokončna VALO:
•
•
•
•
•
•

1 – Polimerizacijska lučka VALO Grand4 – Polnilne baterije ultradent VALO
1 – Polnilec baterij Ultradent VALO z medicinskim napajalnikom 12 VDC
1 – Paket vzorcev pregradnih tulcev VALO Barrier Sleeve
1 – Svetlobni ščit VALO Grand1 – Nosilec za površinsko montažo polimerizacijske lučke z dvojnim lepilnim trakom

Pregled kontrolnikov:

Postopna navodila
Priprava

1) Baterije napolnite pred uporabo polimerizacijske lučke (glejte razdelek Vzdrževanje baterije).
2) Namestite polimerizacijsko lučko v okvir na zobozdravstvenem stroju ali v poseben montažni okvir do naslednje uporabe.
3) Pred vsako uporabo na lučko namestite zaščitni rokav
Higienski zaščitni rokav je prilagojen za polimerizacijsko lučko in ohrani površino lučke čisto. Zaščitni rokav pomaga pri preprečevanju navzkrižne okužbe, preprečevanju lepljenja
kompozitnega materiala na površino leče in lučke, ter preprečuje razbarvanje in rjo od čistilnih sredstev.

Opomba:• Z uporabo higienskega zaščitnega rokava bo svetloba lučke 5-10% šibkejša, vendar se zaradi visoke izhodne moči polimerizacijske lučke kakovost polimerizacije s tem ne spremeni.
• Polimerizacijsko lučko morate po vsakem pacientu očistiti in razkužiti s primernim sredstvom za čiščenje/razkuževanje. Glejte razdelek z naslovom Obdelava.

Gumbi in luči

LegendaGumb za vklop (-e)(Zgoraj ali spodaj)
Indikatorji časaNizka baterija - indikator
Indikator za način/stanje
Gumb za čas / spremembo
načina(Na vrh)

Za vse opisane izdelke pred uporabo natančno preberite vsa navodila in informacije SDS.

Indikacije za uporabo / namen uporabe

Vir osvetlitve za polimerizacijo foto-aktiviranih zobnih obnovitvenih materialov in lepil.

Opozorila in previdnostni ukrepi
Skupina tveganja 2
PREVIDNO ta izdelek seva UV. Izpostavljenost lahko povzroči draženje oči ali kože. Uporabite ustrezno zaščito.
PREVIDNO Možno nevarno optično sevanje tega izdelka. Ne glejte v svetilko. Lahko škoduje očem.
• NE glejte neposredno v svetlobo. Pacient, klinik in pomočniki morajo vedno nositi UV zaščito za oči jantarne barve, ko je v uporabi polimerizacijska svetilka.
• Za preprečitev nevarnosti električnega udara te opreme ni dovoljeno spreminjati. Uporabljajte samo priloženi napajalnik Ultradent VALO in adapterje. Če so te komponente
poškodovane, jih ne uporabljajte in pokličite službo za pomoč strankam Ultradent, da naročite zamenjavo.
• Prenosna komunikacijska oprema RF lahko poslabša delovanje, če se uporablja na razdalji, manjši od 30 cm (12 in.).
• Uporabljajte samo odobreno dodatno opremo, kable in napajalnike, da preprečite nepravilno delovanje, povečane elektromagnetne emisije ali zmanjšano elektromagnetno
odpornost (glejte razdelek o elektromagnetnih emisijah).
• Da bi se izognili nevarnosti električnega požara, povezanega z ravnanjem z baterijami:
⸰ NE mešajte akumulatorskih baterij z baterijami, ki jih ni mogoče polniti, ali drugimi vrstami baterij.
⸰ NE poskušajte polniti baterij, ki jih ni mogoče polniti. Polnjenje baterij za enkratno uporabo lahko povzroči telesne poškodbe ali materialno škodo.
⸰ NE avtoklavirajte in NE pršite baterij, kontaktov baterije, polnilnika ali napajalnika s kakršno koli tekočino. Če se na kontaktih polnilnika baterij pojavi korozija, pokličite servisnoslužbo Ultradent, da naročite zamenjavo.
⸰ NE polnite baterij v bližini vnetljivih materialov.
⸰ NE hranite polnilnika v klinični ambulanti.
• Da bi se izognili nevarnosti poškodb, NE uporabljajte baterij, ki so zarjavele (rjave), vdrte, oddajajo vonj ali tekočino, imajo raztrgan ali manjkajoč ovoj ali so drugače poškodovane. Za
naročilo nadomestnih baterij pokličite službo za pomoč strankam Ultradent.
• Uporabljajte le priporočene baterije. Alternativne baterije lahko povzročijo okvaro.
• Da preprečite tveganje toplotnega draženja ali poškodbe, se izogibajte zaporednim ciklom polimeriziranja in ne izpostavljajte mehkih ustnih tkiv v neposredni bližini za več kot 10
sekund v katerem koli načinu. Če morate polimerizirati dlje časa, uporabite več ciklov polimeriziranja s prekinitvami med cikli ali uporabite izdelek za dvojno polimeriziranja, dapreprečite segrevanje mehkega tkiva.
• Bodite previdni pri zdravljenju pacientov, ki trpijo zaradi neželenih fotobioloških reakcij ali občutljivosti, pacientov, ki se zdravijo s kemoterapijo, ali pacientov, ki se zdravijo z zdravili, ki
povzročajo fotosenzibilizacijo.
• Ta enota je lahko dovzetna za močna magnetna ali statična električna polja, ki bi lahko motila programiranje. Če sumite, da se je to zgodilo, za trenutek izključite enoto in jo nato
ponovno vključite v vtičnico.
• Polimerizacijske lučke NE brišite z jedkimi ali abrazivnimi čistili, avtoklavirajte ali potapljajte v kakršno koli ultrazvočno kopel, razkužilo, čistilno raztopino ali tekočino. Neupoštevanje
priloženih navodil za obdelavo lahko povzroči nedelovanje lučke za polimeriziranje.
232

Da preprečite poškodbe opreme, NE vstavljajte prstov, instrumentov ali drugih predmetov v predal za baterije polimerizacijske lučke.
Da preprečite poškodbe opreme, NE poskušajte očistiti zlatih kontaktov ali katerega koli dela prostora za baterije. Če imate pomisleke, pokličite službo za pomoč strankam Ultradent.
Da bi preprečili tveganje navzkrižne kontaminacije, so pregradni tulci namenjeni uporabi za enega pacienta.
Da zmanjšate tveganje korozije, po uporabi odstranite pregradni tulec.
Da zmanjšate tveganje premalo strjenih smol, ne uporabljajte polimerizacijske luči, če je leča poškodovana.

Namestitev higienskih zaščitnih rokavov:

Fotografija(Spodaj)

•
•
•
•
•

VALO Brezžični svetlobni ščit:VALO akumulatorski ščit za svetlobo je ovalen, lahko ga zavrtite za maksimalno uporabo in ga lahko uporabljate s prozornim zaščitnim rokavom.

Uporaba1) Vsak način napajanja se uporablja za polimerizacijo zobnih materialov s fotografskimi iniciatorji. Za priporočene čase polimerizacije glejte Hitri vodič moči delovanja.
OPOMBA: Polimerizacijska lučka je programirana za zaporedni cikel od Standardne do Visoke Plus in nato Ekstra moči. Na primer, za prehod od Strandardne na Ekstra moč morate
najprej preiti do Visoke moči in šele nato lahko preidete na Ekstra.
2) Polimerizacijska lučka hrani nazadnje uporabljeni časovni interval. Kadar spremenite način ali odstranite baterije, se po privzetem vrne nanj

OperacijaNAČIN ZA POLIMERIZACIJO: Standardni načinČASOVNI INTERVALI: 5, 10, 15, in 20 sekund.
• Polimerizacijska lučka je privzeta za ta način, ko je NAJPREJ vklopljena. Ko je moč standardna, bo indikator Moči delovanja/stanja zasvetil zeleno in prižgali se bodo štirje zeleni
indikatorji časomera.
• Za spremembo časovnega intervala hitro pritisnite gumb za čas/moč.
• Za polimeriziranje pritisnite gumb za vklop. Za zaustavitev polimerizacije preden se izteče nastavljen časovni interval, ponovno pritisnite gumb za vklop

NAČIN ZA POLIMERIZACIJO: Način visoke moči plusČASOVNI INTERVALI: 1, 2, 3, in 4 sekund.
• V standardnem načinu dvakrat pritisnite in držite gumb za čas/moč za dve sekundi, ter spustite Ko je moč Visoka, bo indikator načina/stanja zasvetil oranžno in štirje zeleni indikatorjičasovnega intervala bodo utripali.
• Za spremembo časovnega intervala hitro pritisnite gumb za čas/moč.
• Za polimerizacijo pritisnite gumb za vklop. Za zaustavitev polimerizacije preden se izteče nastavljen časovni interval, ponovno pritisnite gumb za vklop• Za vrnitev v standardni način najprej pritisnite in držite gumb za čas/moč za dve sekundi, nato spustite. S tem boste pognali način Ekstra. Ponovno pritisnite in držite dve sekundi,
nato spustite. Ko je moč standardna, bo indikator načina/stanja zasvetil zeleno in prižgali se bodo štirje zeleni indikatorji časovnega intervala.

NAČIN ZA POLIMERIZACIJO: Način EkstraČASOVNI INTERVAL: Samo 3 sekunde (Opomba: Način Ekstra ima samo dve sekundi varnostnega zamika ob koncu vsakega cikla polimerizacije, kar omeji segrevanje med zaporednimi
polimerizacijami. Pisk ob koncu zakasnitve pomeni, da je lučka pripravljena za nadaljnjo uporabo).
• V standardnem načinu za dve sekundi pritisnite gumb za spremembo časa/moči, spustite, ponovno pritisnite in držite dve sekundi, nato spustite. Ko je moč Ekstra, bo indikator Moči
delovanja/stanja utripal oranžno in bodo utripali trije zeleni indikatorji časovnega intervala.
• Za polimeriziranje pritisnite gumb za vklop. Če želite ustaviti polimerizacijo pred zaključkom časovnega intervala, ponovno pritisnite gumb za vklop.
• Za vrnitev v standardni način najprej pritisnite in držite gumb za čas/moč za dve sekundi, nato spustite. Ko je moč standardna, bo indikator načina/stanja zasvetil zeleno in prižgali sebodo štirje zeleni indikatorji časovnega intervala.

Spanje: Polimerizacijska lučka bo po 60 sekundah nedejavnosti preklopila v način spanja, prikazan s počasnim utripanjem indikatorja načina/statusa. Če lučko dvignete, ali se je
dotaknete, se bo ponovno prebudila in se samodejno vrnila v nazadnje shranjene nastavitve. Da podaljšate življenjsko dobo baterije, lučko pustite pri miru, kadar je ne uporabljate.

Čiščenje

1) Uporabljene zaščitne rokave po vsakem pacientu zavrzite med običajne smeti.
2) Glejte poglavje Obdelava.

Navodila za montažni nosilec

1) Nosilec je treba namestiti na ravno, nemastno površino.
2) Površino očistite z alkoholom.
3) Odlepite zadnji del z lepilnega traku nosilca.
4) Nosilec namestite tako, da bo polimerizacijska lučka obrnjena navzgor. Trdno pritisnite na mesto.

233

Priročnik za hiter način:Standardna moč
1000 mW/cm2

Način

Visoka moč Plus
1600 mW/cm2

Ekstra moč
3200 mW/cm2

Kako polniti baterije:
1) Polnilnik in baterije postavite na mesto, kjer ne bodo prišli v stik s tekočinami ali razkužilom.
2) Baterije vstavite v polnilnik tako, da je pozitivni (+) konec obrnjen proti indikatorskim lučkam na polnilniku.
3) Zelene lučke označujejo, da so baterije pripravljene za uporabo.
4) Baterije se polnijo 1–3 ure. Pustite baterije v polnilniku, dokler niso pripravljene za uporabo.
OPOMBA: Če se rdeča lučka na polnilniku ne spremeni v zeleno, ko se baterije polnijo dlje kot tri ure, je baterija morda izrabljena (poškodovana) in je ni mogoče polniti. Poskusite z novobaterijo ali pokličite službo za pomoč strankam Ultradent, da naročite nov komplet polnilnih baterij.

Gumb za vklop
LED / način delovanjaČasovni gumbiČasovne možnosti

5s

10s

15s

20s

1s

2s

3s

4s

3s Only

Spreminjanje časaHitro pritisnite in spustite gumb za čas, da preklapljate med časovnimi možnostmi.

Spreminjanje načinov

Pritisnite in držite tipko Time za 2 sekundi in jo spustite. Polimerizacijska lučka bo prešla v naslednji način.

LegendaTrdne LED

Načini polimeriziranja

Moč * (mW)

Obsevanost * (mW/cm2)

Skupni čas osvetlitve (sekunde)

Energija** (Juli)

Standard

970

900

5/10/15/20

4,8 J/9,7 J/14,5 J/19,4 J

High Power Plus

1.620

1.500

1/2/3/4

1,6 J/3,2 J/4,8 J/6,5 J

Xtra

2.260

2.100

3

6,8 J

*Nazivni izhod na razdalji 2 mm od konice leče do zgornje površine kompozita.
**Vrednosti v razdelku skupne energije na cikel (džuli) so zaokrožene navzgor na najbližjo desetino.

Opozorilo o ravni moči

Opozorilo o temperaturi

Opozorilo na storitev

LED opozorilo

Zamenjajte baterije

Pustite, da se ohladi

Pokličite servisno službo za popravilo

Pokličite servisno službo za popravilo• Brez zvoka• Utripa, 2 sekundi• Omogoča delovanje• Neprekinjeno 3 piska• Ena hitra bliskavica• Prepoveduje delovanje• Nizka baterija: počasi utripa• Ugasniti: 3 piski, utripajoča• Prepoveduje delovanje• 3 piski• Počasi utripa• Prepoveduje delovanje

Kako zamenjati/vstaviti baterije:
1) Odstranite zadnji pokrov tako, da ga zavrtite za četrt obrata v nasprotni smeri urinega kazalca.
2) Odstranite baterije.
3) Vstavite nove baterije s pozitivno (+) stranjo navzgor.
4) Ponovno pritrdite zadnji pokrov tako, da ga poravnate in nežno potisnete, medtem ko ga obračate v smeri urinega kazalca. Pokrovček bo kliknil, ko bo popolnoma pritrjen.
5) Enota je pripravljena za uporabo.
Nizka raven napolnjenosti baterije: polimerizacijska lučka signalizira uporabniku, da je čas za zamenjavo baterij, ko indikatorska lučka nizke ravni napolnjenosti baterije utripa rdeče. Če
postane napolnjenost baterije prenizka, se zaslišijo trije opozorilni piski, polimerizacijska lučka pa ne dovoli nadaljnjega delovanja, dokler baterije ne napolnite ali vstavite nove baterije.
(Glejte Hitri vodnik za opozorila)

Utripajoče LEDs

Vodnik za hitro sušenje:

Hitri vodič opozoril:Power. OPOMBA: prava številčna vrednost bo popačena zaradi netočnosti običajnih merilnikov svetlobe in prilagojenega paketa LED v polimerizacijski lučkiVzdrževanje baterije
Polnjenje in menjava baterij
Polimerizaijska lučk je opremljena s štirimi litij-železo-fosfatnimi baterijami za polnjenje. Ultradent priporoča uporabo samo polnilnih baterij VALO.

Čas polnjenja in življenjska doba baterije: Življenjska doba popolnoma napolnjene baterije v polimerizacijski lučki je odvisna od načina/časovnega intervala, vrste baterije, količine
uporabe in učinkovitosti LED.
• Priporočeni interval polnjenja: Napolnite po potrebi ali ko se prižge indikator prazne baterije. Pričakovana življenjska doba je odvisna od uporabe.
• Dodatne baterije: Polimerizacijska lučka je opremljena s štirimi baterijami za polnjenje.
Popravilo proizvajalca
1) Popravila lahko izvaja le pooblaščeno servisno osebje. Ultradent mora servisnemu osebju zagotoviti dokumentacijo za izvedbo popravil.
2) Ko pošiljate enote na popravilo, servis ali kalibracije, vedno odstranite baterije iz polimerizacijske lučke in polnilnika. Baterije, polnilnik, adapter in polimerizacijsko lučko ločeno zavijte
v škatlo za vračilo.
3) Baterije pošiljajte v skladu z lokalnimi predpisi.
Garancija
Ultradent Products, Inc. (»Ultradent«) jamči, da bo ta izdelek v obdobju 5 let od datuma nakupa, če se uporablja v skladu z navodili za uporabo, ki so priložena izdelku, (i) v vseh
materialnih vidikih skladen s specifikacijami, navedenimi v Ultradentovi dokumentaciji, ki je priložena izdelku; in (ii) brez napak v materialu in izdelavi.
Ta omejena garancija je neprenosljiva in velja izključno za prvotnega kupca ter ne velja za poznejše lastnike izdelka. Ta omejena garancija ne zajema drugih komponent dodatne opreme,
kot so, vendar ne omejeno na, baterije, polnilniki, adapterji ali prilagodljive leče. Ta omejena garancija je neveljavna, če izdelek odpove ali se poškoduje zaradi malomarnosti, zlorabe,napačne uporabe, nesreče, spreminjanja, poseganja, predelave ali neupoštevanja ustreznih navodil za uporabo. Na primer, ta garancija ne krije izdelka, ki je padel in je poškodovan. Za
izpolnjevanje pogojev za to omejeno garancijo je treba Ultradentu skupaj z okvarjenim izdelkom predložiti dokazilo o nakupu (npr. račun ali podobno dokumentacijo).
Okvarjen izdelek, ki izpolnjuje tukaj določene garancijske pogoje, bo po lastni presoji družbe Ultradent popravljen ali zamenjan. V nobenem primeru odgovornost družbe Ultradentza izdelek ne presega nakupne cene, ki jo plača kupec. Pod nobenim pogojem Ultradent ne odgovarja za kakršno koli posredno, naključno, predvideno, nepredvideno, posebno ali
posledično škodo, ki izhaja iz ali v povezavi z uporabo tega izdelka.

Obdelava

Ta izdelek je namenjen za uporabo z zaščitnim tulcem. Za vzdrževanje in splošno čiščenje glejte zgoraj.

Shranjevanje in odstranjevanjeČe polimerizacijsko lučko shranjujete dlje kot 2 tedna ali jo pakirate za potovanje, baterije vedno odstranite. Če akumulatorji ostanejo v enoti dlje časa, ne da bi jih polnili, lahko izgubijo
funkcionalnost ali ostanejo neuporabni. Baterij ne hranite pri temperaturah nad 40 °C (104 °F) C ali na neposredni sončni svetlobi.

Vzdrževanje

Splošno čiščenje polimerizacijske lučke
Po vsaki uporabi navlažite gazo ali mehko krpo z odobrenim razkužilom za površine in obrišite površino in lečo. Nepooblaščena čistila lahko poškodujejo polimerizacijsko lučko.
SPREJEMLJIVA ČISTILA:• 70% izopropilni alkohol• 70% etanol

Splošno čiščenje svetlobnega ščita:Brezžični svetlobni ščit VALO očistite s poljubnim razkužilom za površine. NE avtoklavirajte.
PopraviloVzdrževanje, ki ga izvaja uporabnik
1) Uporabite pregradni tulec, da se zobni kompozit ne prime na površino leče. Če je potrebno, uporabite zobni instrument iz plastike ali nerjavečega jekla, da skrbno odstranite
morebitni prilepljeni kompozit. Ne uporabljajte orodja, ki bi poškodovala lečo.
2) Merilniki svetlobe se med sabo močno razlikujejo in so zasnovani za posebne konice svetlobnega vodnika in leče. Ultradent priporoča redno preverjanje delovanja v načinu Standard
234

Skladiščenje in shranjevanje:• Temperatura: + 10 ° C do + 40 ° C (+ 50 ° F do + 104 ° F)
• Relativna vlažnost: 10% do 95%• Tlak okolice: 500 hPa do 1060 hPa
Pri odlaganju elektronskih odpadkov (tj. polimerizacijske lučke, polnilniki, baterije in napajalniki) upoštevajte lokalne smernice za odpadke in recikliranje.

Tehnični vidiki
DodatkiPostavka

Informacije o CEBrezžični svetlobni ščit VALO

235

Postavka

Informacije o CE

Tehnični podatki / podatki

Učinkoviti kompozitni trakovi za utrjevanje valovnih dolžin:Polnilnik VALOPASOVI VALOVNIH DOLŽIN UČINKOVITEGA UTRJEVANJA KOMPOZITA(Uporabno območje valovne dolžine: 385-515 nm)
SPEKTRALNA MOČ SEVANJA (mW/nm)

Baterije VALO

Učinkovito nižje
pas valovne dolžine
395-415 nm

Efektivni pas zgornjevalovne dolžine
440-480 nm

XtraHigh PowerStandard

WAVEL ENGTH (nm)Lastnost

Informacije / specifikacije

Objektiv

Premer 11,7 mm

Razpon valovnihdolžin• Uporabljivo območje valovnih dolžin: 385 - 515 nm• Najvišje valovne dolžine: 395 - 415 nm in 440 - 480 nm
Tabela nominalnega izstopnega sevanja
Merilni instrument

Tabela intenzivnosti
svetlobe236‡ Gigahertzov analizator spektra
Izhod

Skupna moč

Odprtina merilnika

15 mm

15 mm

Standardna moč(±10%)

900 mW/cm²

970 mW

Visoka moč Plus(±10%)

1500 mW/cm²

1615 mW

Ekstra moč (±10%)2100 mW/cm²

Nominalno obsevanje se bo razlikovalo glede
na zmogljivost instrumenta, metodo merjenja
in postavitev svetlobe.
‡ Nominalno obsevanje je v skladu z ISO
10650, če se meri z analizatorjem spektra
Gigahertz-Optik.

2260 mW

Polimerizacijska
lučka Imenitna
VALO

Ocene: IEC 60601-1 (Varnost), IEC 60601-1-2
(EMC)

Teža:
• Z baterijami: 6 oz. (190 gramov)• Brez baterij: 5 oz. (150 gramov)
Dimenzija: (8 x 1,28 x 1,06) palcev (203 x 32,5 x 27) mm

Napajalnik
polnilnika

Izhod - 12VDC pri 500mA
Vhod - 100VAC do 240VAC
Ultradent P / N 5930 Napajalnik za polnilnik
VALO z mednarodnimi vtičniki

Ocene: IEC 60601-1 (Varnost)Dolžina kabla - 1,8 metra
Napajalnik VALO za polnilnik je napajalnik medicinske kakovosti II. razreda
in zagotavlja izolacijo od moči MAINS

Polnilnik VALO

Polnilec za pametne akumulatorske litijeve železove fosfate 3.6VDC VALO:• Samodejni izklop ob polni napolnjenosti• Samodejno zaznavanje okvarjenih baterij• Zaščite: Toplotna, prenapetostna, proti kratkemu stiku, povratna polarnost⸰ Rdeča LED - polnjenje
⸰ Zelena LED - prazna ali popolnoma napolnjena
⸰ Izklop LED - kratek stik• Čas polnjenja: 1 - 3 ure
Ocena: CE, OEEO
237

Razne informacije

Lastnost

Informacije / specifikacije

Baterije VALO

Polnilne: Varna kemija litij-železo-fosfat (LiFePO4) RCR123A• Delovna napetost: 3,2 VDC, 400 mAh (potrebna vatna ura)
Ocene: UL 62133-2, CE, RoHS, WEEE
Celice so ocenjene za >2000 ciklov polnjenja/praznjenja, preden dosežejo 70 % svoje zmogljivosti. Priporočljivo je zamenjatibaterije vsakih 3–5 let, vendar se življenjska doba lahko razlikuje glede na uporabo. Po potrebi zamenjajte.

Pogoji delovanja

Temperatura: + 10 ° C do + 32 ° C (+ 50 ° F do + 90 ° F)
Relativna vlažnost: 10% do 95%
Tlak okolice: 700 hPa do 1060 hPa

Delovni cikel:

Polimerizacijska lučka je zasnovana za kratkotrajno delovanje. Pri najvišji temperaturi okolja (32° C) 1 minuto ON na vzvratno
kolesarjenje, 30 minut OFF (obdobje hlajenja).

Odpravljanje težav
Če spodaj predlagane rešitve ne odpravijo težave, pokličite Ultradent na številko 800.552.5512. Zunaj Združenih držav Amerike pokličite svojega distributerja ali
prodajalca zob.
TežavaSvetloba se ne bo vklopila

Izjavo o elektromagnetnih emisijah
Polimerizacijska lučka je namenjena uporabi v spodaj navedenem elektromagnetnem okolju. Stranka ali uporabnik mora zagotoviti, da se uporablja v takšnem okolju.
Preskus emisij

Skladnost

Elektromagnetno okolje - vodenje

RF emisije
CISPR 11

Skupina 1

Polimerizacijska lučka uporablja baterije in ne vpliva na EMI, RF ali prenapetost.

RF emisija
CISPR 11

Razred B

Polimerizacijska lučka uporablja električno in elektromagnetno energijo samo za svoje notranje
funkcije. Zato so vse radiofrekvenčne emisije zelo nizke in verjetno ne bodo povzročile motenj vbližnji elektronski opremi.

Harmonične emisije
IEC 61000-3-2

N/A

Preskušanje nihanja harmonskih emisij in napetosti se ne uporablja za polimerizacijsko lučko, ker
je na baterijsko napajanje.

Nihanja napetosti / emisije
flikerja IEC 61000-3-3

N/A

Polimerizacijska lučka je primerna za uporabo v vseh obratih, vključno z domačimi in tistimi, ki so
neposredno priključeni na javno napajalno omrežje, ki oskrbuje zgradbe za domačo uporabo.

Možne rešitve
1) Pritisnite gumb za čas / način ali gumb za vklop, da se prebudite iz načina varčevanja z energijo.
2) Preverite rdeč indikator stanja napolnjenosti baterije.
3) Preverite, ali so sveže baterije pravilno vstavljene v napravo.
4) Če utripata rdeča in rumena opozorilna svetleča dioda, to pomeni, da je svetloba za strjevanje dosegla varnostno mejo
notranje temperature. Počakajte, da se svetloba strjevanja ohladi 10 minut, ali pa uporabite hladno mokro brisačo, dase naprava hitro ohladi.
5) Če rdeča opozorilna LED utripa in piska neprekinjeno, pokličite servisno službo Ultradent za popravilo.

Svetloba ne ostane na želenem času1) Preverite način in časovne luči za pravilen časovni vnos.
2) Preverite indikator prazne baterije za stanje napolnjenosti baterije.
3) Preverite, ali so sveže baterije pravilno vstavljene v napravo.

Svetloba ne strdi pravilno smole

1) Preverite leče za preostale strjene smole / kompozite.
2) Z ustrezno zaščito pred oranžnimi UV očmi preverite, ali LED luči delujejo.
3) Preverite nivo moči z merilnikom svetlobe. Če uporabljate merilnik svetlobe, Ultradent priporoča preverjanje
polimerizacijske lučke v standardnem načinu.
OPOMBA: Zaradi netočnosti običajnih merilnikov svetlobe in prilagojenega sklopa LED, katere uporablja polimerizacijska
lučka, lahko prikazane številčne vrednosti odstopajo od dejanskih. Merilniki svetlobe se zelo razlikujejo in so zasnovani za
posebne svetlobne vodnike in leče.
4) Preverite datum izteka roka trajanja na suhi smoli.
5) Zagotovite, da se v skladu s priporočili proizvajalca upoštevajo pravilne tehnike.

Izjavo o elektromagnetni imuniteti
Polimerizacijska lučka je namenjena uporabi v spodaj navedenem elektromagnetnem okolju. Stranka ali uporabnik mora zagotoviti, da se uporablja v takšnem okolju.
Test IMMUNITY

IEC 60601 testni nivo

Raven skladnosti

Navodila za elektromagnetno okolje

Elektrostatična razelektritev (ESD)

Kontakt ± 8 kV
± 15 kV zraka

Kontakt ± 8 kV
± 15 kV zraka

Fizično okolje mora biti omejeno na naslednje:
1) IP koda: IP20
2) Ne potapljajte v tekočino.
3) Ne uporabljajte okoli vnetljivih plinov. Enota jene-APG in Non-AP.
4) Območje vlažnosti za shranjevanje: 10% - 95%
5) Razpon temperatur skladiščenja: 10 ° C - 40 ° C

Električno hitro prehodno /počenoIEC 61000-4-4± 2 kV za napajalne vode± 1 kV za vhodne / izhodne linije± 2 kV za napajalne vode
Opomba 1: Svetloba za sušenje
nima I / 0 vrat

Polimerizacijska lučka se napaja iz akumulatorja in sene more povezati z napajanjem AC MAINS.

Surge
IEC 61000-4-5± 1 kV± 2 kV vod na zemljo± 1 kV± 2 kV vod na zemljo

Napetost, padci, kratki spoji,
prekinitve in spremembe na
vhodnih napajalnih vodih<5% U(> 95% dip v ciklu U za 0,5)<5% U(> 95% dip v ciklu U za 0,5)40% U
(60% dip v U za 5 ciklov)

40% U(60% dip v U za 5 ciklov)

IEC 61000-4-2

Baterije se ne polnijo

1) Prepričajte se, da so baterije pravilno vstavljene v polnilnik in da se baterije polnijo 1-3 ure.
2) Če se rdeče luči na polnilniku ne spremenijo v zeleno, pokličite službo za stranke Ultradent in naročite nadomestnebaterije in / ali polnilnik.
3) Če na polnilniku ni vidnih zelenih ali rdečih luči, pokličite servisno službo za stranke Ultradent in naročite ali zamenjajte
polnilnik in / ali napajalnik.

Polnilnik ne polni baterij

1) Prepričajte se, da je polnilnik priključen in da je napajalnik priključen na delujočo vtičnico.
2) Če zelena ali rdeča lučka na polnilniku ni vidna, pokličite servis za stranke Ultradent za nov polnilnik in / ali napajalnik.

70% U(30% dip v U za 25 ciklov)

70% U(30% dip v U za 25 ciklov)

Način ali časovne intervale ni mogoče
spremeniti

1) Tipke Time / Mode in Power držite pritisnjene, dokler niz piskov ne pokaže, da je lučka za sušenje odklenjena.

<5% U(> 95% dip v U za 5 s)

<5% U(> 95% dip v U za 5 s)
Opomba 2: Sam se opomore

30 A/m30 A/mIEC 61000-4-11

Pogonska frekvenca(50/60 Hz) magnetno polje
IEC 61000-4-8

Ker je polimerizacijska lučka napajana baterijsko, ni
podvržena električnim prehodom, prenapetostim,
napetostnim padcem, kratkim spojem, prekinitvam ali
spremembam napajanja AC MAINS.
Dodatni polnilnik akumulatorjev je lahko predmet
zgoraj navedenega, vendar je ločen od in ne kritičen
za delovanje VALO svetilke za sušenje brez kablov.
Če napetost akumulatorja na svetlobi strjevanja
pade na 4VDC, naprava ne bo omogočala delovanja.
Brezžična polimerizacijska lučka VALO se bo izklopila.
Ko vstavite nove baterije in se ponovno vzpostavijo
ustrezne ravni moči, se bo polimerizacijska lučka
ponovno zagnala in se vrnila v isto stanje pred izgubo
energije. Polimerizacijska lučka se bo samodejno
obnovila v primeru izpada električne energije.
Magnetna polja napetostne frekvence morajo
biti na ravneh, značilnih za tipično lokacijo vtipičnem, stanovanjskem, domačem zdravstvenem,
komercialnem, bolnišničnem ali vojaškem okolju.

Opomba 1: Polimerizacijska lučka ni opremljena z nobenimi priključki ali dostopnimi linijami V / I.

238

239

Izjava o elektromagnetni imuniteti za sisteme neživljenjske podpore

Izjava o smernicah in izdelavi za priporočene razdalje med prenosno in mobilno RF komunikacijsko opremo in brezžično svetlobo VALO

Polimerizacijska lučka je namenjena uporabi v spodaj navedenem elektromagnetnem okolju. Stranka ali uporabnik mora zagotoviti, da se uporablja v takšnem okolju.
Test IMMUNITY

IEC 60601 testni nivo

Raven skladnosti

Prevajanje RF

3 Vrms3 VrmsIEC 61000-4-6

150 kHz do 80 MHz

150 kHz do 80 MHz

Navodila za elektromagnetno okolje
Prenosne in mobilne opreme za komunikacijo RF ne uporabljajte preblizu nobenega dela
VALO, vključno s kabli. Razdalja ne sme biti manjša od priporočene ločitvene razdalje,izračunane na osnovi enačbe glede na frekvenco oddajnika.
Priporočena razdalja:

80 MHz do 800 MHz

800 MHz do 2.5 GHz
Sevani RF

3V/m

3V/m

P je največja nazivna izhodna moč oddajnika v vatih (W) glede na proizvajalca oddajnika in
d je priporočena razdalja v metrih (m).
Jakosti polja iz fiksnih oddajnikov RF, kot jih določa elektromagnetni pregled mestaa, mora
biti pod mero skladnosti pri vsakem razponu frekvenceb.

IEC 61000-4-3

80 MHz do 2,5 GHz

80 MHz do 2,5 GHz

V bližini opreme, označene za naslednjim simbolom

lahko pride do interference:

Svetloba za sušenje je namenjena za uporabo v elektromagnetnem okolju, v katerem se nadzirajo radiofrekvenčne motnje. Uporabnik svetleče svetlobe lahko
pomaga preprečiti elektromagnetne motnje z vzdrževanjem minimalne razdalje med prenosno in mobilno RF komunikacijsko opremo (oddajniki) in svetlobo za
strjevanje, kot je priporočeno spodaj, glede na največjo izhodno moč komunikacijske opreme.
Ločilna razdalja glede na frekvenco oddajnika(metrov)
Nazivna največja izhodna moč
oddajnika(P v vatih)

150 kHz – 80 MHz

80 MHz – 800 MHz

800 MHz – 2,5 GHz

0,01

0,12 metra

0,035 metra

0,07 metra

0,1

0,37 metra

0,11 metra

0,22 metra

1

1,7 metra

0,35 metra

0,7 metra

10

3,7 metra

1,11 metra

2,22 metra

100

11,7 metra

3,5 metra

7,0 metra

Polimerizacijska lučka je bila testirana v skladu z IEC 60601-1-2: 2014 in prešla pod sevano poljsko jakostjo 10 V / m med 80 MHz do 2,5 GHz. Vrednost 3Vrms ustreza
V1 in vrednost 10 V / m ustreza E1 v zgornjih formulah.
Za oddajnike z največjo izhodno močjo, ki niso navedeni zgoraj, je priporočljivo razdaljo d v metrih (m) mogoče oceniti z enačbo, ki velja za frekvenco oddajnika, pričemer je P največja nazivna izhodna moč oddajnika v vatih (W) glede na proizvajalca oddajnika.
OPOMBA 1: Pri 80 MHz in 800 MHz se uporablja ločilna razdalja za višje frekvenčno območje.
OPOMBA 2 Te smernice morda ne veljajo v vseh primerih. Na elektromagnetno širjenje vplivajo absorpcija in refleksija struktur, predmetov in ljudi.

OPOMBA 1: Pri 80 MHz in 800 MHz velja višje frekvenčno območje.
OPOMBA 2 Te smernice morda ne veljajo v vseh primerih. Na elektromagnetno širjenje vplivajo absorpcija in refleksija struktur, predmetov in ljudi.
aJakosti polja iz fiksnih oddajnikov, kot so bazne postaje za radijske (mobilne / brezžične) telefone in kopenske mobilne radijske aparate, amaterski radijski
sprejemnik, radijsko in radijsko oddajanje AM in FM ter TV oddajanje, ni mogoče teoretično natančno predvideti. Za oceno elektromagnetnega okolja zaradi fiksnih
RF oddajnikov je treba upoštevati elektromagnetni pregled mesta. Če izmerjena moč polja na mestu, kjer uporabljate VALO presega zgoraj navedeno veljavno
skladnost RF, morate brezžično lučko VALO opazovati tako, da zagotovite normalno delovanje. Če opazite nenormalno delovanje, bodo morda potrebni dodatni
ukrepi, kot je preusmeritev ali premestitev strjevanja.
b
V frekvenčnem območju od 150 kHz do 80 MHz morajo biti jakosti polja manjše od 3 V / m.

240

241

VALO™ GRAND

Полимеризационная Лампа

RU

Описание продукта:Благодаря широкополосному спектру отверждающий светильник VALO Grand предназначен для полимеризации всех продуктов светоотверждения в диапазоне длин волн 385-515
нм в соответствии со стандартом ISO 10650. В лампе для отверждения VALO Grand используются ультратонкие аккумуляторные батареи VALO и зарядное устройство для них. Лампадля отверждения предназначена для установки в стандартный кронштейн стоматологического аппарата или может быть установлена на заказ с помощью кронштейна, входящего вкомплект поставки.

VALO Гранд Компоненты изделия:
•
•
•
•
•
•

Фото

Кнопки и индикаторы

Примечание:• Чтобы предотвратить перекрестное загрязнение и предотвратить прилипание стоматологического композитного материала к поверхности линзы и корпусу палочки, при каждом
использовании необходимо надевать защитную пленку, одобренную компанией Ultradent. Защитные чехлы предназначены для одноразового использования.

Щиток беспроводной лампы VALO:

Условные
обозначения

Кнопка питания
(Верхний или Нижний)Индикаторы времени
Индикатор разряда элементов питания
Световой индикатор режима/состояния
Кнопка смены времени/режима

(Нижний)(Верхний)Перед использованием внимательно прочитайте и уясните инструкции для всех перечисленных продуктов, а также всю информацию в паспорте безопасности.

Показания к использованию / Назначение

Данная лампа является источником света для отверждения светочувствительных восстановительных стоматологических материалов и адгезивов.

Меры предосторожности и предупреждения
Группа риска 2ВНИМАНИЕ! этот продукт излучает ультрафиолетовые лучи. В результате воздействия может возникнуть раздражение глаз или кожи. Используйте специальную защиту.
ВНИМАНИЕ! Возможно опасное оптическое излучение, испускаемое этим продуктом. Не смотрите на работающую лампу. Может быть вредно для глаз.
• НЕ смотрите прямо на источник света. Пациент, врач-клиницист и ассистенты всегда должны носить УФ-защиту для глаз янтарного цвета при использовании лечебного света.
• Во избежание риска поражения электрическим током запрещается вносить изменения в данное оборудование. Используйте только входящий в комплект поставки блок питания
Ultradent VALO и штекерные адаптеры. Если эти компоненты повреждены, не используйте их и позвоните в службу поддержки клиентов Ultradent, чтобы заказать замену.
• Портативное оборудование радиочастотной связи может ухудшить производительность при использовании ближе, чем на 30 см (12 дюймов).
• Используйте только разрешенные аксессуары, кабели и источники питания для предотвращения неправильной эксплуатации, повышенного электромагнитного излучения илиснижения помехозащищенности от электромагнитных помех (см. раздел "Электромагнитное излучение").
• Во избежание риска возгорания электропроводки при обращении с аккумуляторными батареями:⸰ НЕ смешивайте перезаряжаемые батареи с неперезаряжаемыми батареями или батареями других типов.
⸰ НЕ пытайтесь заряжать неперезаряжаемые батареи. Зарядка неперезаряжаемых аккумуляторных батарей может привести к травмам или повреждению имущества.
⸰ НЕ подвергайте автоклавированию и не обрызгивайте батареи, контакты аккумулятора, зарядное устройство или адаптер переменного тока какой-либо жидкостью. Еслина контактах зарядного устройства появилась коррозия, позвоните в службу поддержки Ultradent, чтобы заказать замену.
⸰ НЕ заряжайте аккумуляторы вблизи легковоспламеняющихся материалов.
⸰ НЕ храните зарядное устройство в операционной.
• Во избежание риска получения травмы ЗАПРЕЩАЕТСЯ использовать батареи с коррозией (ржавчиной), вмятинами, запахом или жидкостью, из надорванной упаковки или безупаковки, а также с другими повреждениями. Позвоните в службу поддержки клиентов Ultradent, чтобы заказать замену батареек.
• Используйте только рекомендованные аккумуляторные батареи. Альтернативные аккумуляторы могут привести к неисправности.
• Чтобы предотвратить риск термического раздражения или травмы, избегайте последовательных циклов отверждения и не подвергайте мягкие ткани полости рта воздействию внепосредственной близости более 10 секунд в любом режиме. Если требуется более длительное время отверждения, используйте несколько циклов отверждения с перерывами
между циклами или используйте средство двойного отверждения, чтобы избежать нагревания мягких тканей.
• Соблюдайте осторожность при лечении пациентов, страдающих неблагоприятными фотобиологическими реакциями или повышенной чувствительностью, пациентов,проходящих курс химиотерапии, или пациентов, получающих фотосенсибилизирующие препараты.
• Данное устройство может быть подвержено воздействию сильных магнитных или статических электрических полей, которые могут привести к сбою программирования. Привозникновении подозрений отключите устройство от сети на некоторое время, а затем снова включите его в розетку.
• НЕ протирайте отверждающий светильник едкими или абразивными чистящими средствами, не подвергайте автоклавированию и не погружайте в какие-либо ультразвуковые

242

Пошаговые инструкции
Подготовка

1) Зарядите аккумуляторы перед использованием лампы для отверждения (см. раздел Обслуживание аккумуляторов).
2) Перед каждым использованием надевайте новый защитный чехол на лампу, минимизируя таким образом образование неровностей на линзе для достижения наилучших
результатов.

1 – Отверждающий светильник VALO Grand4 – Разрядные аккумуляторные батареи VALO Ultradent
1 – Зарядное устройство Ultradent VALO с адаптером питания переменного тока медицинского класса 12 В постоянного тока1 – Пакет образцов защитной гильзы VALO
1 – VALO Grand световой щит
1 – Монтажный кронштейн для отверждения легкой поверхности с двойной клейкой лентой

Обзор элементов управления:ванны, дезинфицирующие средства, чистящие растворы или жидкости. Несоблюдение прилагаемых технологически инструкций может привести к неработоспособности
устройства.
• Во избежание повреждения оборудования НЕ вставляйте пальцы, инструменты или другие предметы в батарейный отсек лампы для отверждения.
• Чтобы избежать повреждения оборудования, не пытайтесь чистить золотые контакты или какую-либо часть батарейного отсека. В случае возникновения проблем позвоните вслужбу поддержки клиентов Ultradent.
• Чтобы предотвратить риск перекрестного заражения, барьерные рукава предназначены для одноразового использования пациентом.
• Чтобы снизить риск коррозии, снимите защитную втулку после использования.
• Чтобы снизить риск неполного отверждения смол, не используйте отверждающий свет, если линза повреждена.

Световой щиток беспроводной лампы VALO имеет овальную форму, может поворачиваться для максимально удобного использования, а также может использоваться спрозрачными защитными чехлами.

Использование

1) Каждый режим питания используется для отверждения стоматологических материалов с фотоинициаторами. См. Краткое руководство по режимам для рекомендуемого времени
отверждения.
ПРИМЕЧАНИЕ: Полимеризационная лампа запрограммирована на циклическое и последовательное переключение между режимами стандартной, высокой плюс и максимальной
мощности. Например, чтобы изменить мощность со стандартной на максимальную, необходимо
2) Полимеризационная лампа всегда запоминает последний использованный временной интервал и режим и по умолчанию возвращается к этим настройкам при изменении
режима или извлечении элементов питания.

Эксплуатация
РЕЖИМ ОТВЕРЖДЕНИЯ: РЕЖИМ СТАНДАРТНОЙ МОЩНОСТИ

ВРЕМЕННЫЕ ИНТЕРВАЛЫ: 5, 10, 15, и20 секунд.
• Полимеризационная лампа по умолчанию переключается в этот режим при ПЕРВОНАЧАЛЬНОМ включении ее питания. Индикатор режима/состояния загорится зеленым
цветом, а четыре зеленых индикатора времени будут гореть, указывая на режим стандартной мощности.
• Чтобы изменить временной интервал, быстро нажмите кнопку смены времени/режима.
• Нажмите кнопку питания, чтобы выполнить процедуру отверждения. Чтобы остановить процедуру отверждения до окончания временного интервала, снова нажмите кнопку
питания.

РЕЖИМ ОТВЕРЖДЕНИЯ: РЕЖИМ ВЫСОКОЙ МОЩНОСТИ

ВРЕМЕННЫЕ ИНТЕРВАЛЫ: 1, 2, 3, и4 секунды.
• Находясь в стандартном режиме, нажмите и удерживайте кнопку смены времени/режима в течение 2 секунд и отпустите. Индикатор режима/состояния загорится оранжевым
цветом, а четыре зеленых индикатора времени будут гореть и мигать, указывая на режим высокой мощности.
• Чтобы изменить временной интервал, быстро нажмите кнопку смены времени/режима.
• Нажмите кнопку питания, чтобы выполнить процедуру отверждения. Чтобы остановить процедуру отверждения до окончания временного интервала, снова нажмите кнопку
питания.
• Чтобы вернуться к стандартному режиму мощности, нажмите и удерживайте кнопку смены времени/режима в течение 2 секунд и отпустите, чтобы перейти в режим
максимальной мощности. Нажмите и удерживайте в течение 2 секунд, а затем отпустите. Индикатор режима/состояния загорится зеленым цветом, а четыре зеленых индикатора
времени будут гореть, указывая на режим стандартной мощности.

РЕЖИМ ОТВЕРЖДЕНИЯ: РЕЖИМ МАКСИМАЛЬНОЙ МОЩНОСТИ

ВРЕМЕННОЙ ИНТЕРВАЛ: только 3 секунды (примечание: Режим максимальной мощности имеет 2-ух секундные перерывы по завершении каждого цикла отверждения с целью
обеспечения безопасности, чтобы предотвратить нагрев в процессе последовательного отверждения. В конце каждого перерыва звуковой сигнал сообщает о том, что устройство
готово к продолжению работы).
• Находясь в стандартном режиме, нажмите кнопку смены времени/режима в течение 2 секунд, отпустите, затем снова нажмите и удерживайте кнопку в течение 2 секунд,снова отпустите. Индикатор режима/состояния будет мигать оранжевым цветом, а три зеленых индикатора времени будут гореть и мигать, указывая на режим максимальной
мощности.
• Нажмите кнопку питания, чтобы выполнить процедуру отверждения. Чтобы остановить процедуру отверждения до окончания временного интервала, снова нажмите кнопку
питания.
• Чтобы вернуться в стандартный режим, нажмите и удерживайте кнопку смены времени/режима в течение 2 секунд, затем отпустите. Индикатор режима/состояния загорится
зеленым цветом, а зеленые индикаторы времени будут гореть, указывая на режим стандартной мощности.

Режим ожидания: Через 60 секунд бездействия полимеризационная лампа переходит в режим ожидания, о чем свидетельствует медленно мигающий зеленый индикатор режима/состояния. Если взять полимеризационную лампу в руки или коснуться ее, она перейдет в рабочий режим и автоматически возвратится к последней использованной настройке.
Чтобы продлить срок службы батареи, не дотрагивайтесь до полимеризационной лампы, когда она не используется.

Очистка

1) Выбрасывайте использованные защитные чехлы после каждого пациента.
2) См. Раздел «Обработка».

Инструкции по установке держателя

1) Держатель должен быть размещен на плоской обезжиренной поверхности.
2) Очистите поверхность при помощи медицинского спирта.
3) Снимите защитный слой с липкой ленты на держателе.
4) Разместите держатель таким образом, чтобы лампа поднималась вверх при снятии. Плотно прижмите держатель на месте крепления.

Краткое руководство по режимам:243

Стандартная мощность
1000 mW/cm2

РежимВысокая мощность плюс1600 mW/cm2

Максимальная мощность
3200 mW/cm2

Кнопка питания
Светодиоды режима/времени

Кнопка времени

Опции времени

5c

10c

15c

20c

1c

2c

3c

4c

Только 3cЧтобы изменить времяДля изменения временных интервалов быстро нажимайте и отпускайте кнопку времени.

Чтобы изменить режим

Нажмите и удерживайте кнопку времени в течение 2 секунд, а затем отпустите. Полимеризационная лампа переключитсяна следующий режим.

Условные обозначения

Светодиоды горят

Режим отверждения

Мощность* (МВт)Освещенность* (МВт/см2)Общее время экспозиции (сек)Энергия**(Джоули)Стандарт

970

900

5/10/15/20

4,8Дж/9,7Дж/14,5
Дж/19,4ДжВысокая мощность
Плюс

1,620

1,500

1/2/3/4

1,6Дж/3,2Дж/4,8
Дж/6,5Дж

2,260

2,100

3

6,8Дж

*Номинальная мощность на расстоянии 2 мм от кончика линзы до верхней поверхности композита.
**Значения в разделе общая энергия за цикл (джоули) округлены до ближайшей десятой.

Краткое руководство по предупреждающим сигналам:Предупреждение об уровне
мощности
Замените элементы питания
• Разряжены элементы питания:медленное мигание
• Выключение: 3 звуковых
сигнала, мигание
• Работа лампы блокированаКак заряжать аккумуляторы:1) Поместите зарядное устройство и аккумуляторы в такое место, где они не будут соприкасаться с жидкостями или дезинфицирующими спреями.
2) Вставьте батарейки в зарядное устройство положительным (+) концом, направленным в сторону индикаторных лампочек на зарядном устройстве.
3) Зеленые индикаторы указывают на то, что батареи готовы к использованию.
4) Зарядка аккумуляторов займет 1-3 часа. Оставьте батарейки в зарядном устройстве до готовности к использованию.
ПРИМЕЧАНИЕ: Если красный индикатор на зарядном устройстве не загорается зеленым, когда батареи заряжаются более трех часов, возможно, батарея разряжена (повреждена) ине может быть заряжена. Попробуйте новую батарею или позвоните в службу поддержки Ultradent, чтобы заказать новый комплект аккумуляторных батарей.
Как заменить/вставить батарейки:1) Снимите заднюю крышку, повернув против часовой стрелки на четверть оборота.
2) Извлеките батарейки.
3) Сначала вставьте новые батарейки положительной (+) стороной.
4) Установите заднюю крышку на место, выровняв и слегка надавив, одновременно поворачивая по часовой стрелке. Колпачок защелкнется, когда будет полностью закреплен.
5) Устройство готово к использованию.

Светодиоды мигают

Краткое руководство по отверждению:XtraГенеральная чистка светового экрана:Очистите беспроводной световой экран VALO, используя любое дезинфицирующее средство для поверхностей. НЕ подвергайте автоклавированию.
Ремонт
Техническое обслуживание, выполняемое пользователем
1) Используйте защитную втулку, чтобы предотвратить прилипание стоматологического композита к поверхности линзы. При необходимости используйте стоматологический
инструмент из пластика или нержавеющей стали, чтобы аккуратно удалить прилипший композит. Не используйте инструменты, которые могут повредить линзу.
2) Измерители освещенности сильно различаются и предназначены для конкретных наконечников световодов и линз. Ultradent рекомендует регулярно проверять выходную
мощность в стандартном режиме питания. ПРИМЕЧАНИЕ: истинный числовой результат будет искажен из-за неточности обычных измерителей освещенности и специального
светодиодного блока в отверждающем светильнике
Техническое обслуживание батареи
Зарядка и замена аккумуляторов
Лампа для отверждения поставляется с 4 перезаряжаемыми литий-железо-фосфатными батареями. Ultradent рекомендует использовать только аккумуляторные батареи VALO.

Предупреждение о высокой
температуре

Предупреждение об обслуживании

Предупреждение о светодиодах

Дайте возможность охладиться

Позвоните в службу поддержки
клиентов для ремонта

Позвоните в службу поддержки
клиентов для ремонта

• 3 звуковых сигнала
• Медленное мигание
• Работа лампы блокирована

• Без звука
• Мигает каждые 2 секунды
• Работа лампы разрешена

• 3 непрерывных звуковых
сигнала
• Одиночное быстрое мигание
• Работа лампы блокирована

Разряженные батареи: Индикатор отверждения сигнализирует пользователю о том, что пришло время заменить батареи, когда индикатор разряженной батареи мигает красным.
Если заряд батареи станет слишком низким, раздастся звуковой сигнал с 3 звуковыми сигналами, а индикатор отверждения не позволит продолжить работу до тех пор, пока батареине будут заряжены или не будут вставлены новые. (См. Краткое руководство по предупреждению)Время зарядки и срок службы батареи: Время работы полностью заряженной батареи в режиме отверждения зависит от режима/временного интервала, типа батареи, объема
использования и эффективности светодиода.
• Рекомендуемый интервал подзарядки: Подзаряжайте по мере необходимости или при включении индикатора низкого заряда батареи. Ожидаемая продолжительность жизни
зависит от использования.
• Дополнительные батарейки: Лампа для отверждения поставляется с 4 перезаряжаемыми батарейками.
Ремонт от производителя
1) Ремонт должен выполняться только авторизованным сервисным персоналом. Ultradent должна предоставить обслуживающему персоналу документацию для выполнения
ремонта.
2) При отправке устройств на ремонт, сервисное обслуживание или калибровку всегда извлекайте батарейки из лампы для отверждения и зарядного устройства. Упакуйте батарейки,зарядное устройство, адаптер и лампу для отверждения отдельно в коробку для возврата.
3) Отправляйте аккумуляторы в соответствии с местными правилами.
Гарантия
Ultradent Products, Inc. (“Ultradent”) гарантирует, что данное изделие в течение 5 лет с даты покупки при эксплуатации в соответствии с инструкциями по эксплуатации,прилагаемыми к изделию, (i) будет соответствовать во всех существенных отношениях техническим характеристикам, изложенным в документации Ultradent, прилагаемой кизделию; и (ii) не должно быть дефектов материала и изготовления.
Данная ограниченная гарантия не подлежит передаче и распространяется исключительно на первоначального покупателя и не распространяется на последующих владельцев
изделия. Данная ограниченная гарантия не распространяется на любые другие вспомогательные компоненты, такие как батареи, зарядные устройства, адаптеры или адаптивные
линзы, но не ограничивается ими. Данная ограниченная гарантия утрачивает силу, если изделие выходит из строя или повреждено в результате халатности, неправильного
обращения, ненадлежащего использования, несчастного случая, модификации, фальсификации, переделки или несоблюдения применимых инструкций по использованию. Толькодля примера, данная гарантия не распространяется на упавшее и поврежденное изделие. Чтобы подпадать под действие данной ограниченной гарантии, Ultradent необходимо
предоставить подтверждение покупки (например, товарный чек или аналогичную документацию) вместе с дефектным изделием.
Дефектный продукт, соответствующий условиям гарантии, изложенным в настоящем документе, будет, по собственному усмотрению Ultradent, отремонтирован или заменен. Ни вкоем случае ответственность Ultradent за продукт не должна превышать покупную цену, уплаченную покупателем. Ни при каких обстоятельствах Ultradent не несет ответственностиза любые косвенные, случайные, предвиденные, непредвиденные, особые или косвенные убытки, возникающие в результате использования данного продукта или в связи с ним.

Обращение

Данное изделие предназначено для использования с защитным рукавом. Инструкции по техническому обслуживанию и генеральной уборке приведены выше.

Хранение и утилизация
Техническое обслуживание

Генеральная очистка отверждающего светильника
После каждого использования смочите марлю или мягкую ткань специальным дезинфицирующим средством и протрите поверхность и линзу. Несоответствующие чистящие
средства могут привести к повреждению фотополимерной лампы.
ДОПУСТИМЫЕ ЧИСТЯЩИЕ СРЕДСТВА:•		
70 %-ный изопропиловый спирт
•		
70 %-ный этанол

244

Если вы храните полимеризационную лампу более 2 недель или упаковываете ее для поездки, всегда извлекайте элементы питания. Если элементы питания оставить в лампе надлительный период времени без подзарядки, они могут перестать работать или их невозможно будет зарядить. •Не храните элементы питания при температуре свыше 400°C (104°F)или под воздействием прямых солнечных лучей.
Условия хранения и транспортировки полимеризационной лампы:• Температура: от + 10 °C до + 40 °C (от + 50 °F до + 104 °F)• Относительная влажность: От 10% до 95%• Давление внешней среды: 500 гПа до 1060 гПаПри утилизации электронных отходов (т.е. полимеризационных ламп, зарядных устройств, аккумуляторов и источников питания) соблюдайте местные правила утилизации ипереработки.

245

Технические рекомендации
Вспомогательные принадлежности

Атрибут

Предмет

Информация / Спецификация
Таблица сравнения номинальных значений радиационного выхода

Информация СЕИзмеритель-ныйприбор

Щиток беспроводной лампы VALO

‡ Гигагерцовый анализатор спектра
Cветимость

Макси-мальная
мощ-ность

Диафрагма метра

15 мм

15 мм

Стандартная
мощность (±10%)1000 мВт/см²

750 мВт

Элементы питания VALO

Максималь-наямощность (±10%)2100 мВт/см²

1570 мВт

Технические данные и информация

Режим максимальной квадратной
мощности (±10%)2100 мВт/см²

1570 мВт

Таблица интенсивности излучения

Зарядное устройство VALO

СПЕКТРАЛЬНАЯ МОЩНОСТЬ ИЗЛУЧЕНИЯ (МВт/нм)Рабочая полоса волнового диапазона для отверждения композитов:ЭФФЕКТИВНЫЕ ДИАПАЗОНЫ ДЛИН ВОЛН ДЛЯ ОТВЕРЖДЕНИЯ
КОМПОЗИТОВ
(Используемый диапазон длин волн: 385-515 нм)Эффективный
более низкий
диапазон длин волн395-415 нмЭффективный
верхний диапазон
длин волн
440-480 нмДЛИНА ВОЛНЫ (нм)Атрибут

Информация / Спецификация

ЛинзаДиаметр 11,7 ммВолновой
диапазон

• Используемый волновой диапазон: 385 – 515 нм• Пиковые длины волн: 395 – 415 нм и 440 – 480 нмРежим XtraРежим высокой мощности

Мощность излучения будет варьироватьсяв зависимости от возможностей прибора,метода измерения и расположения
источника света.
‡ Мощность излучения соответствует
стандарту ISO 10650 при измерении спомощью анализатора спектра GigahertzOptik.

Вес:• С элементами питания: 6 унций. (190 грамм)• Без элементов питания: 5 унций. (150 грамм)Размеры: (8 x 1,28 x 1,06) дюймов, (203 x 32,5 x 27) ммПолимеризационная лампа
VALO ГРАНД

Рейтинг: МЭК 60601-1 (безопасность), МЭК60601-1-2 (ЭМС)Блок питания
зарядного
устройства

Рейтинг: МЭК 60601-1 (безопасность)На выходее – 12 В постоянного тока при 500 мАДлина шнура – 6 футов (1,8 метра)На входе – от 100 В до 240 В переменного токаЭлектропитание зарядного устройства VALO является источником
Ultradent P/N 5930 VALO Блок питания спитания медицинского назначения класса II и обеспечивает изоляцию
универсальными штекерамиот основного электропитания.

Зарядное
устройство VALO

Зарядное устройство для литий-железо-фосфатных аккумуляторов VALO на 3,6 В пост. тока с интеллектуальным
управлением:• Автоматическое выключение после полной зарядки аккумуляторов
• Автоматическое выявление дефектных аккумуляторов
• Защита: Тепловая, от перезарядки, короткого замыкания, обратной полярности⸰ Красный светодиод – идет зарядка⸰ Зеленый светодиод – разряженный или полностью заряженный аккумулятор⸰ Светодиод не горит – короткое замыкание
• Время зарядки: 1–3 часаРейтинг: CE, WEEEСтандартный режим

Перезаряжаемый: Безопасный химический литий-железо-фосфат (LiFePO4) RCR123A•
Рабочее напряжение: 3,2 В постоянного тока, 400 мАч (требуется ватт-час)Элементы питания Рейтинги: UL 62133-2, CE, RoHS, WEEE
VALO
Элементы рассчитаны на более чем 2000 циклов зарядки/разрядки, прежде чем их емкость достигнет 70%.
Рекомендуется заменять батарейки каждые 3-5 лет, но срок службы может варьироваться в зависимости отиспользования. Заменяйте по мере необходимости.
Температура: от + 10 °C до + 32 °C (от + 50 °F до + 90 °F)Условия эксплуатаОтносительная влажность: От 10% до 95%цииДавление внешней среды: От 700 гПа до 1060 гПаКоэффи-циент
заполнения

Полимеризационная лампа предназначена для кратковременной работы. При максимальной температуре окружающего
воздуха (32 °C) лампа может быть включена 1 минуту на непрерывном цикле и 30 минут выключена (период
охлаждения).

Поиск неисправностей

246

247

Если предложенные ниже решения не помогают устранить неполадки, пожалуйста, позвоните в компанию Ultradent по номеру 800.552.5512. За пределамиСША звоните торговому представителю Ultradent или агенту по продаже стоматологических изделий.
Неисправность

Possible Solutions1) Нажмите кнопку смены времени/режима или кнопку питания, чтобы вывести устройство из энергосберегающего
режима.
2) Проверьте состояние элементов питания по красному индикатору разряда.
3) Убедитесь, что в устройство установлены заряженные элементы питания с соблюдением полярности.
4) Если мигают красный и желтый предупреждающие светодиоды, это означает, что в полимеризационной лампе
достигнуто предельное значение внутренней температуры. Дайте полимеризационной лампе остыть в течение 10минут или остудите лампу при помощи прохладного влажного полотенца.
5) Если мигает красный предупреждающий индикатор и непрерывно подается звуковой сигнал, позвоните в отдел
обслуживания потребителей Ultradent, чтобы отремонтировать изделие.

Лампа не включается

Полимеризационная лампа предназначена для использования в электромагнитных условиях, описанных ниже. Клиент/пользователь должен убедиться, чтоприбор используется в данных условиях.
Испытание на устойчивость

Тестовый уровень в соответствии
Уровень соответствия
с IEC 60601

Устойчивость к электростатическим разрядам

± 8 кВ при контактном разряде
± 15 кВ воздух

± 8 кВ при контактном разряде
± 15 кВ воздух

Физическая среда должна быть ограничена
следующим:1) IP-код: IP202) Не погружайте в жидкость.
3) Не используйте вблизи
легковоспламеняющегося газа. Прибор не
APG/AP.
4) Допустимая влажность при хранении:10% – 95%5) Температурный диапазон: 10 °C – 40 °C
Полимеризационная лампа питается отаккумулятора и не может подключаться к сетипеременного тока.

IEC 61000-4-2Регламент по электромагнитному излучению

Лампа выключается раньше
требуемого времени

1) Проверьте индикаторы режима и настроек времени на предмет введения правильной настройки времени.
2) Проверьте состояние элементов питания по индикатору разряда.
3) Убедитесь, что в устройство установленные заряженные элементы питания с соблюдением полярности.

Лампа отверждает смолы неполностью

1) Проверьте линзу на наличие отвержденных остатков смолы/композитов на ее поверхности.
2) Используя соответствующую защиту глаз янтарного цвета от УФ-излучения, убедитесь в том, что светодиодные
индикаторы работают.
3) Проверьте уровень мощности фотометром. Если используется фотометр, Ultradent рекомендует проводить
проверку полимеризационной лампы в стандартном режиме.
ПРИМЕЧАНИЕ: Действительное числовое значение мощности на выходе будет иметь отклонения в связи с неточностью
обычных Секторальной и набора уникальных светодиодов, используемых в полимеризационной лампе. Секторальной
сильно отличаются друг от друга и разрабатываются для наконечников и линз конкретных световодных систем.
4) Проверьте срок годности Секторальной.
5) Убедитесь, что соблюдается правильная методика, рекомендованная производителем. .

Быстрые электрические
± 2 кВ для линий
переходные процессы/ всплески электропередачи
IEC 61000-4-4± 1 кВ для входных/выходных
линий± 2 кВ для линий
электропередачи
Примечание 1:Полимеризационная лампа неимеет портов ввода/вывода

Устойчивость к микросе-кундным ± 1 кВ от линии к линии
импульсным помехам большой ± 2 кВ от линии к земле
энергии
IEC 61000-4-5± 1 кВ от линии к линии
± 2 кВ от линии к земле

Поскольку полимеризационная лампа питаетсяот аккумулятора, она не подвержена воздействию
электрических переходных процессов, скачков
напряжения, провалов напряжения, коротких
замыканий, перерывов или колебаний питания
переменного тока.

Аккумуляторы не заряжаются

1) Убедитесь, что аккумуляторы установлены в зарядное устройство с соблюдением полярности и дайте имзарядиться в течение 1-3 часов.
2) Если цвет световых индикаторов на зарядном устройстве не переключаются с красного на зеленый, позвоните вотдел обслуживания клиентов Ultradent и закажите новые аккумуляторы и/или зарядное устройство.
3) Если не горят ни красные, ни зеленые световые индикаторы на зарядном устройстве, позвоните в отдел
обслуживания клиентов Ultradent, чтобы заменить зарядное устройство и/или сетевой блок питания.

Напряжение, провалы,короткие замыкания, перебои иизменения на входных линиях
питания

<5% U(> 95% провал от U для 0,5цикла)<5% U(> 95% провал от U для 0,5цикла)

40% U(60% провал от U для 5 циклов)40% U(60% провал от U для 5 циклов)Вышеперечисленное может применяться кзарядному устройству, но так как оно является
отдельным устройством, данные условия некритичны для работы полимеризационной лампы.

70% U(30% провал от U для 25 циклов)70% U(30% провал от U для 25 циклов)30 А/м

30 А/м

Зарядное устройство не заряжает
аккумуляторы

1) Убедитесь, что зарядное устройство подключено, а блок питания от сети переменного тока включен в работающую
розетку.
2) Если на зарядном устройстве не горят ни зеленые, ни красные световые индикаторы, позвоните в отдел
обслуживания клиентов Ultradent и закажите новое зарядное устройство и/или сетевой блок питания.

Невозможно изменить режим илиинтервалы времени

1) Удерживайте нажатыми кнопки время/режим и мощность до тех пор, пока серия звуковых сигналов не покажет,что свет отверждения не разблокирован.

Различная информация
Руководство и декларация производителя по электромагнитным излучениям
Полимеризационная лампа предназначена для использования в электромагнитных условиях, описанных ниже. Клиент/пользователь должен убедиться, чтоприбор используется в данных условиях.

248

Руководство и декларация производителя по электромагнитной стойкости

Испытание на излучение

Соответ-ствие

Электромагнитная среда – руководство

Радиочистотное излучениеCISPR 11Группа 1В полимеризационной лампе используются аккумуляторы, которые не подвергаются влиянию
электромагнитных, радиочастотных помех или подавлению импульсных перенапряжений.

Радиочастотное излучениеCISPR 11Класс ВГармоническое излучение
IEC 61000-3-2Неприме-нимоПолимеризационная лампа использует электрическую и электромагнитную энергию толькодля работы внутренних функций. Таким образом, уровень любых РЧ-излучений чрезвычайно
низок и в большинстве случаев не способен вызывать помехи в работе расположенного
поблизости электронного оборудования.

Колебания напряжения /мерцающее излучение IEC
61000-3-3Неприме-нимоГармоническое излучение и колебания напряжения не применимы к работе
полимеризационной лампы, так как она работает на аккумуляторах.

Если напряжение аккумулятора
полимеризационной лампы упадет до 4 Впостоянного тока, устройство не будет работать.
Беспроводная полимеризационная лампа VALO
выключится. Когда будут вставлены новые
аккумуляторы и соответствующие уровни мощности
<5% U
<5% U
(> 95% провал от U в течение 5 с) (> 95% провал от U в течение 5 с) будут восстановлены, полимеризационная лампа
перезапустится и вернется в тот же режим, чтоПримечание 2:и до потери питания. В случае потери питания,Самовосстанавли-вается
полимеризационная лампа восстановится
самостоятельно.

IEC 61000-4-11

Частота импульса:(50/60 Гц) магнитного поляIEC 61000-4-8Магнитные поля с частотой питающей сети должны
находиться на уровнях, характерных для типичных
мест в обычной жилой, домашней, медицинской,коммерческой, больничной или военной
обстановке.

Примечание 1: Полимеризационная лампа не оборудована портами или любыми другими доступными линиями ввода/вывода.

Руководство и декларация изготовителя по электромагнитной стойкости для систем жизнеобеспечения
Полимеризационная лампа предназначена для использования в электромагнитных условиях, описанных ниже. Клиент/пользователь должен убедиться, чтоприбор используется в данных условиях.
Испытание наустойчивость

Тестовый уровень всоответствии с IEC
60601

Уровень соответствия Регламент по электромагнитному излучению

Полимеризационная лампа подходит для использования в любых учреждениях, в томчисле в бытовых и подключенных напрямую к низковольтной электрической сети общего
пользования, которая питает здания и используется для бытовых целей.

249

Руководство и декларация изготовителя по электромагнитной стойкости для систем жизнеобеспечения

Кондуктивная
радиочастота

3 В ср. квадр.

3 В ср. квадр.

IEC 61000-4-6

От 150 кГц до 80 МГц

От 150 кГц до 80 МГцИзлучаемая
радиочастота

3 В/м

3 В/м

Портативное и мобильное радиочастотное оборудование не должно быть ближе кполимеризационной лампе (включая ее кабели), чем рекомендуемое расстояние,рассчитанное из уравнения, применимого к частоте передатчика.
Рекомендованное расстояние:От 80 МГц до 800 МГцРуководство и декларация производителя для рекомендованных расстояний между переносным и мобильным оборудованием радиосвязи и беспроводной
полимеризационной лампой VALO
Полимеризационная лампа была протестирована в соответствии с IEC 60601-1-2:2014 и прошла тестирование под воздействием напряженности поля 10
В/м от 80 кГц до 2,5 ГГц. Значение 3Vrms (среднеквадратичное напряжение сигнала) соответствует V1, а значение 10 В/м соответствует E1 в приведенных
выше формулах.
Для передатчиков, рассчитанных на максимальную выходную мощность, не указанную выше, рекомендуемое расстояние разделения d в метрах (м) можно
оценить, используя уравнение, применимое к частоте передатчика, где P – максимальная выходная мощность передатчика в ваттах (Вт) в соответствии спроизводителем передатчика.
ПРИМЕЧАНИЕ 1. При 80 МГц и 800 МГц применяется расстояние разделения для более высокого частотного диапазона.
ПРИМЕЧАНИЕ 2. Данные указания могут относиться не ко всем ситуациям. На распространение электромагнитных волн влияет поглощение и отражение отконструкций, предметов и людей.

От 800 МГц до 2,5 ГГц
P – максимальная выходная мощность передатчика в ваттах (Вт) согласно
производителю передатчика, а d – рекомендуемый пространственный разнос вметрах (м).
Значения напряженности электромагнитного поля стационарных РЧ-передатчиков,как установлено в электромагнитной съемке объектаa, должны быть ниже уровня
соответствия для каждого диапазона частотb.
IEC 61000-4-3

От 80 МГц до 2,5 ГГцОт 80 МГц до 2,5 ГГц Помехи могут возникать вблизи оборудования, отмеченного следующим символом:ПРИМЕЧАНИЕ 1. При 80 МГц и 800 МГц применяется более высокий частотный диапазон.
ПРИМЕЧАНИЕ 2. Данные указания могут относиться не ко всем ситуациям. На распространение электромагнитных волн влияет поглощение и отражение отконструкций, предметов и людей.
aЗначения напряженности электромагнитного поля стационарных передатчиков, таких как базовые станции для радио- (сотовых/беспроводных) телефонови наземных подвижных радиостанций, аппаратура радиолюбительской связи, а также теле- и радиовещательные станции диапазонов AM и FM, невозможно
предсказать теоретически с достаточной степенью точности. Для оценки электромагнитной обстановки, связанной с фиксированными радиопередатчиками,следует провести электромагнитное обследование площадки. Если измеренная напряженность электромагнитного поля в месте эксплуатации
полимеризационной лампы превышает указанный выше применимый уровень соответствия, необходимо произвести осмотр беспроводной ламы VALO сцелью проверки ее нормальной работоспособности. При наличии признаков нарушения эксплуатационных характеристик может потребоваться принять
дополнительные меры, например изменить направление или местоположение полимеризационной лампы.
bВ диапазоне частот от 150 кГц до 80 МГц значения напряженности электромагнитного поля должны быть меньше 3 В/м.
Руководство и декларация производителя для рекомендованных расстояний между переносным и мобильным оборудованием радиосвязи и беспроводной
полимеризационной лампой VALO
Полимеризационная лампа предназначена для использования в электромагнитных условиях, в которой контролируются излучаемые радиопомехи.
Исходя из рекомендаций ниже, соответствующих максимальной выходной мощностью коммуникационного оборудования, клиент/пользователь
полимеризационной лампы может помочь предотвратить воздействие электромагнитных помех, поддерживая минимальное расстояние между портативными мобильным радиокоммуникационным оборудованием (передатчиками) и полимеризационной лампой.

Расчетная максимальная выходная
мощность
передатчика
(P в Ваттах)250Расстояние разделения по частоте передатчика
(метры)150 кГц – 80 МГц

80 МГц – 800 МГц

800 МГц – 2,5 ГГц

0,01

0,12 метра

0,035 метра

0,07 метра

0,10,37 метра

0,11 метра

0,22 метра

1

1,7 метра

0,35 метра

0,7 метра

10

3,7 метра

1,11 метра

2,22 метра

100

11,7 метра

3,5 метра

7,0 метра251

VALO™ 그랜드

KO경화 라이트사용자 프로필광대역 스펙트럼을 갖춘 VALO Grand 광중합기는 ISO 10650에 따라 385-515nm의 파장 범위에서 모든 광중합 제품을 중합하도록 설계되었습니다. VALO Grand광중합기는 Ultradent VALO 충전식 배터리와 배터리 충전기를 사용합니다. 광중합기는 표준 치과 장치 브래킷에 놓이도록 설계되거나 키트에 포함된 브래킷을사용하여 맞춤 장착할 수 있습니다.

VALO 그랜드 제품의 구성:
•
•
•
•
•
•

1 – VALO Grand 광중합기4 – Ultradent VALO 충전식 배터리
1 – 의료용 12VDC AC 전원 어댑터가 있는 Ultradent VALO 배터리 충전기
1 – VALO 배리어 슬리브 샘플 팩
1 – VALO Grand 라이트 쉴드
1 – 양면 접착 테이프가 있는 광중합기 표면 장착 브래킷제어장치의 개요:사진• 교차 오염을 방지하고 치과용 복합 재료가 렌즈와 봉 본체의 표면에 달라붙는 것을 방지하려면 사용할 때마다 VALO 광중합기 위에 Ultradent 승인 배리어 슬리브를사용해야 합니다. 배리어 슬리브는 단일 환자용으로 제작되었습니다.

용도버튼 및 라이트전원 버튼(상단 또는 하단)범례모드/상태 표시등타이밍/모드 변경 버튼(상단)설명된 모든 제품에 대해, 사용 전에 모든 지침과 물질안전보건자료(SDS) 정보를 주의 깊게 읽고 이해하십시오.

사용 방법 / 용도광 활성화 치과용 수복재 및 접착제 경화용 조명 원천.

1) 각 전원 모드는 광 개시제로 치과 재료를 경화하는 데 사용됩니다. 권장 경화 시간은 빠른 모드 안내를 참조하십시오.
주: 경화 라이트는 표준전력에서 고전력, 엑스트라 전력 모드까지 순차적으로 전환되도록 구성되어 있습니다. 예를 들어, 표준전력 모드에서 엑스트라 전력 모드로변경하려면, 고전력 모드를 거친 다음 엑스트라 전력 모드로 넘어가야 합니다.
2) 경화 라이트는 가장 최근에 사용된 타이밍 간격과 모드를 저장하며, 모드가 변경되거나 배터리가 제거될 때마다 기본값으로 돌아갑니다.

작동경화 모드: 표준전력 모드시간 간격: 5, 10, 15, 20초.
• 경화라이트는 처음으로 전원을 켤 때, 이 모드로 기본설정이 되어 있습니다. '모드/상태' 표시등이 녹색이 되고 4개의 녹색 '시간 표시등'이 켜지고 표준전력 모드가표시됩니다.
• 시간 간격을 변경하려면, '시간/모드' 버튼을 빠르게 누릅니다.
• 전원 버튼을 눌러 경화합니다. 하나의 시간 간격을 완료하기 전에 경화를 중지하려면, 전원 단추를 다시 누르십시오.

경화 모드: 고전력 플러스 모드시간 간격: 1, 2, 3, 4초.
• 표준전력 모드에서, '시간/모드 변경' 버튼을 2초 동안 눌렀다가 놓습니다. '모드/상태' 표시등이 주황색으로 켜지고, 4 개의 녹색 '시간 표시등'이 켜져 깜빡거림으로써고전력 모드임을 나타냅니다.
• 시간 간격을 변경하려면, '시간/모드' 버튼을 빠르게 누릅니다.
• 어느 '전원 버튼'이든 눌러 경화하십시오. 하나의 시간 간격을 완료하기 전에 경화를 중지하려면, 어느 전원 버튼이든 다시 누릅니다.
• 표준전력 모드로 돌아가려면, '시간/모드 변경' 버튼을 2초 동안 눌렀다 놓으면 엑스트라(Xtra) 전력 모드로 전환됩니다. 다시 2초 동안 눌렀다가 뗍니다. 모드/상태표시등이 녹색으로 되고, 4개의 녹색 '시간 표시등'이 켜져 표준 모드임을 나타냅니다.

경화 모드: 엑스트라 전력 모드위험군 2주의 이 제품에서 방사되는 자외선. 노출 시 눈이나 피부에 자극을 줄 수 있음. 적절한 차폐를 사용하십시오.

주의 이 제품에서 방사될 가능성이 있는 유해한 광학 방사선. 작동 중인 램프를 쳐다보지 마십시오. 눈에 해를 끼칠 수 있습니다.

• 나오는 조명을 직접 쳐다보지 마십시오. 환자, 임상의, 조수는 광중합기를 사용할 때 항상 호박색 UV 보안경을 착용해야 합니다.
• 감전의 위험을 방지하기 위해 이 장비를 개조할 수 없습니다. 포함된 Ultradent VALO 전원 공급 장치 및 플러그 어댑터만 사용하십시오. 이러한 구성 요소가 손상된경우 사용하지 말고 Ultradent 고객 서비스에 전화하여 교체품을 주문하십시오.
• 휴대용 RF 통신 장비를 30cm(12인치) 이내에서 사용하면 성능이 저하될 수 있습니다.
• 부적절한 작동, 전자기 방출 증가 또는 전자기 내성 감소를 방지하기 위해 허가된 액세서리, 케이블 및 전원 공급 장치만 사용하십시오(전자기 방출 섹션 참조).
• 배터리 취급과 관련된 전기 화재의 위험을 방지하려면:⸰ 충전식 배터리를 비충전식 배터리 또는 기타 유형의 배터리와 혼용하지 마십시오.
⸰ 비충전식 배터리를 충전하지 마십시오. 비충전식 배터리를 충전하면 부상이나 재산 피해가 발생할 수 있습니다.
⸰ 배터리, 배터리 접점, 충전기 또는 AC 전원 어댑터에 어떤 종류의 액체도 오토클레이브하거나 분사하지 마십시오. 배터리 충전기 접점에 부식이 나타나면 Ultradent고객 서비스에 전화하여 교체품을 주문하십시오.
⸰ 가연성 물질 주변에서 배터리를 충전하지 마십시오.
⸰ 충전기를 임상 수술실에 보관하지 마십시오.
• 부상의 위험을 방지하려면 부식(녹)되거나, 찌그러지거나, 냄새나 액체가 나거나, 포장이 찢어지거나 누락되었거나, 손상된 배터리를 사용하지 마십시오. Ultradent고객 서비스에 전화하여 교체용 배터리를 주문하십시오.
• 권장 배터리만 사용하십시오. 비정품 배터리는 오작동을 일으킬 수 있습니다.
• 열 자극이나 부상의 위험을 방지하려면 연속 경화 사이클을 피하고, 어떤 모드에서든 구강 연조직을 10초 이상 가까이 노출시키지 마십시오. 더 긴 경화 시간이 필요한경우 사이클 사이에 휴식 기간이 있는 다중 경화 사이클을 사용하거나 이중 경화 제품을 사용하여 연조직에 열을 가하는 것을 방지하십시오.
• 광생물학적 부작용이나 민감성으로 고통받는 환자, 화학 요법 치료를 받고 있는 환자 또는 감광제로 치료를 받는 환자를 치료할 때는 주의하십시오.
• 본 기기는 강한 자기장 또는 정전기장에 취약하여 그러한 요소들이 기기의 프로그래밍을 방해할 수 있습니다. 이런 일이 발생한 것으로 의심되면 장치의 플러그를잠시 뽑았다가 콘센트에 다시 꽂으십시오.
• 부식성 또는 연마성 세제, 오토클레이브로 광중합기를 닦아내거나 모든 종류의 초음파 수조, 소독제, 세척액 또는 액체에 담그지 마십시오. 포함된 처리 지침을 따르지않으면 광중합기가 작동하지 않을 수 있습니다.
• 장비 손상을 방지하려면 광중합기의 배터리 칸에 손가락, 기구 또는 기타 물체를 삽입하지 마십시오.
• 장비 손상을 방지하려면 도금 접점이나 배터리 구획의 일부를 청소하지 마십시오. 우려 사항이 있는 경우 Ultradent 고객 서비스에 전화하십시오.

2521) 광중합기를 사용하기 전에 배터리를 충전하십시오(배터리 유지 관리 섹션 참조).
2) 매번 사용하기 전에 광중합기 위에 새 배리어 슬리브를 씌우고 최상의 결과를 위해 렌즈 슬리브의 주름을 최소화하십시오.

주:

VALO 코드리스 라이트 실드:배터리 부족 – 표시등경고 및 사전 주의사항단계별 지침
준비

VALO 코드리스 라이트 실드는 타원형이며, 최대한의 사용을 위해 회전 가능하며, 투명한 배리어 슬리브와 함께 사용할 수 있습니다.

타이밍 표시등(하단)• 교차 오염의 위험을 방지하기 위해 배리어 슬리브는 1인 환자 전용입니다.
• 부식될 위험을 줄이려면 사용 후 배리어 슬리브를 제거하십시오.
• 경화 불량 수지의 위험을 줄이려면 렌즈가 손상된 경우 광중합기를 사용하지 마십시오.

시간 간격: 3초만 해당 (주: 엑스트라 전력 모드는 연속 경화 중의 가열을 제한하기 위해 각 경화 주기의 끝에서 2초의 안전 지연 시간을 두고 있습니다. 지연이 끝나면신호음이 울려 장치가 계속 사용될 준비가 되었음을 알립니다).
• 표준전력 모드에서, '시간/모드 변경' 버튼을 2초 동안 눌렀다가 떼고, 다시 2초 동안 눌렀다가 뗍니다. '모드/상태' 표시등이 주황색으로 켜져 깜빡거리고, 3개의 녹색 '시간 표시등'이 켜져 깜빡거림으로써 엑스트라 전력 모드임을 나타냅니다.
• 전원 버튼을 눌러 경화합니다. 하나의 시간 간격을 완료하기 전에 경화를 중지하려면, 전원 버튼을 다시 누릅니다.
• 표준전력 모드로 돌아가려면, '시간/모드' 버튼을 2초 동안 눌렀다가 뗍니다. 모드/상태 표시등이 녹색이 되고, 녹색의 시간 표시등이 켜짐으로써 표준전력 모드임을나타냅니다.
수면 모드: 경화 라이트는 60초 동안 사용하지 않으면 모드/상태 표시등이 천천히 깜박이면서 '수면 모드'로 들어갑니다. 장치를 들거나 만지면 경화 라이트가 켜지며자동으로, 마지막으로 사용한 설정으로 되돌아갑니다. 배터리 수명을 최대화하려면 사용하지 않을 때 경화 라이트를 그대로 두십시오.

세척1) 각 환자에게 사용한 후, 사용된 배리어 슬리브를 표준 폐기물로서 버리십시오.
2) 처리 섹션을 참조하세요.

브래킷 장착 지침1) 브래킷은 기름이 없는 평평한 표면에 설치해야 합니다.
2) 소독용 알코올로 표면을 세척합니다.
3) 브래킷의 접착 테이프를 벗겨 내십시오.
4) 제거할 때 경화 라이트가 위쪽을 향해 들리도록 브래킷 위치를 잡습니다. 단단히 눌러 제자리에 들어가게 합니다.

253

빠른 모드 가이드:표준전력1000 mW/cm2

모드

전원 버튼

고전력 플러스
1600 mW/cm2

엑스트라 전력
3200 mW/cm2

모드/타이밍 LED
시간 버튼
시간 옵션

5초

시간을 변경하려면

10초

15초

20초

1초

2초

3초

'시간 버튼'을 눌렀다 빠르게 놓으면 시간 옵션이 순환합니다.

모드를 변경하려면

솔리드형 LED빠른 경화 가이드:점멸형 LED경화 모드

출력* (mW)조도* (mW/cm2)

총 노출 시간(초)고출력 플러스

1,620

1,500

1/2/3/4

표준

970

Xtra

900

2,260

2,100

온도 경고

• 배터리량 부족: 천천히깜박거림• 끔: 신호음 3 회, 깜박거림• 작동 금지냉각 허용• 신호음 3회• 천천히 깜박거림• 작동 금지

에너지** (줄)4.8J/9.7J/14.5J/19.4J

3

6.8J

서비스 경고수리를 위해 고객 서비스부에 연락• 사운드 없음• 깜박거림, 2초마다• 작동 허용배터리 충전 방법:1) 충전기와 배터리는 액체나 소독 스프레이와 접촉하지 않는 곳에 두십시오.
2) 양극(+) 끝이 충전기의 표시등을 향하도록 배터리를 충전기에 삽입합니다.
3) 녹색 표시등은 배터리를 사용할 준비가 되었음을 나타냅니다.
4) 배터리를 충전하는 데 1-3시간이 걸립니다. 사용할 준비가 될 때까지 배터리를 충전기에 그대로 연결해 두십시오.
알림: 배터리를 3시간 이상 충전했을 때 충전기의 빨간색 표시등이 녹색으로 바뀌지 않으면 배터리 성능이 저하(손상)되어 충전할 수 없는 것일 수 있습니다. 새 배터리를사용하거나 Ultradent 고객 서비스에 전화하여 새 충전식 배터리 세트를 주문하십시오.
배터리 교체/삽입 방법:1) 시계 반대 방향으로 1/4 바퀴 돌려 백 캡을 제거합니다.
2) 배터리를 제거하십시오.
3) 새 배터리의 양극(+) 쪽을 먼저 삽입하십시오.
4) 시계 방향으로 돌리면서 정렬하고 부드럽게 밀어 백 캡을 다시 부착합니다. 캡이 완전히 부착되면 딸깍 소리가 납니다.
5) 장치를 사용할 준비가 되었습니다.

배터리 부족: 배터리 부족 표시등이 빨간색으로 깜박일 때 광중합기는 사용자에게 배터리 교체 시기임을 알립니다. 배터리 충전량이 너무 낮아지면 경고음이 3번 울리고배터리를 재충전하거나 새 배터리를 삽입할 때까지 광중합기가 더 이상 작동하지 않습니다. (빠른 경고 가이드 참조)

5/10/15/20*렌즈 끝에서 복합재의 상단 표면까지 2mm 거리에서의 공칭 출력.
**사이클당 총 에너지(줄) 섹션의 값은 가장 가까운 10분의 1로 반올림됩니다.

전원 수준 경고배터리 교체

3초만 해당

'시간' 버튼을 2초 동안 눌렀다가 뗍니다. 경화 라이트는 다음 모드로 순환합니다.

범례빠른 경고 안내서:4초권장합니다. 알림: 실제 숫자 출력은 광중합기의 일반 조도계와 맞춤형 LED 팩의 부정확성으로 인해 왜곡됩니다.
배터리 유지 관리배터리 충전 및 교체광중합기는 4개의 충전식 인산철 리튬 배터리와 함께 제공됩니다. Ultradent는 VALO 충전식 배터리만 사용할 것을 권장합니다.

1.6J/3.2J/4.8J/6.5J

LED 경고수리를 위해 고객 서비스부에 연락• 연속 3회 신호음• 한번의 빠른 깜빡임• 작동 금지충전 시간 및 배터리 수명: 광중합기의 완전히 충전된 배터리 수명은 모드/시간 간격, 배터리 유형, 사용량 및 LED 효율에 따라 다릅니다.
• 권장 충전 간격: 필요에 따라 또는 배터리 부족 표시등이 켜질 때 충전하십시오. 기대 수명은 사용에 따라 다릅니다.
• 추가 배터리: 광중합기는 4개의 충전식 배터리와 함께 제공됩니다.

제조업체 수리
1) 수리는 공인 서비스 직원만 수행해야 합니다. Ultradent는 서비스 담당자에게 수리를 수행하기 위한 문서를 제공하는 것입니다.
2) 수리, 서비스 또는 교정을 위해 장치를 보낼 때는 항상 광중합기 및 충전기에서 배터리를 제거하십시오. 배터리, 충전기, 어댑터 및 광중합기를 반환 상자에 별도로포장하십시오.
3) 현지 규정에 따라 배터리를 배송하십시오.
보증
Ultradent Products, Inc.(이하 "Ultradent")는 본 제품이 제품에 포함된 사용 설명서에 따라 작동할 경우 구매일로부터 5년 동안 (i) 제품과 함께 제공된 Ultradent의문서에 명시된 사양에 모든 중요한 측면에서 부합하며 (ii) 재료 및 제조상의 결함이 없음을 보증합니다.

이 제한 보증은 양도할 수 없으며 최초 구매자에게만 적용되며 이후 제품 소유자에게는 적용되지 않습니다. 이 제한 보증은 배터리, 충전기 또는 어댑터, 어댑티브 렌즈와같은 기타 액세서리 구성품에는 적용되지 않습니다. 본 제한 보증은 과실, 남용, 오용, 사고, 개조, 변조, 변경 또는 해당 사용 지침을 따르지 않아 제품이 고장 나거나손상된 경우 무효화됩니다. 예를 들어, 제품을 떨어뜨려서 손상된 제품은 이 보증이 적용되지 않습니다. 이 제한 보증을 적용받으려면 구매 증빙 자료(예: 판매 영수증 또는이와 유사한 문서)를 결함이 있는 제품과 함께 Ultradent에 제출해야 합니다.
여기에 명시된 보증 조건을 충족하는 결함이 있는 제품은 Ultradent의 단독 재량에 따라 수리 또는 교체됩니다. 어떠한 경우에도 제품에 대한 Ultradent의 책임은구매자가 지불한 구매 가격을 초과하지 않습니다. 어떠한 경우에도 Ultradent는 이 제품의 사용으로 인해 또는 이와 관련하여 발생하는 간접적, 부수적, 예견된, 예상치못한, 특별 또는 결과적 손해에 대해 책임을 지지 않습니다.

처리이 제품은 배리어 슬리브와 함께 사용하도록 만들어졌습니다. 유지 관리 및 일반 청소에 대해서는 위를 참조하십시오.

보관 및 처분

경화라이트를 2주 이상 보관하거나 포장하여 운반할 경우에는, 항상 배터리를 분리하십시오. 배터리를 재충전하지 않고 장시간 기기에 방치하면 작동하지 않거나 충전이되지 않을 수 있습니다. 40°C (104°F) 이상의 고온이나 직사광선이 내리 쬐는 곳에 배터리를 보관하지 마십시오.

유지보수광중합기의 일반 청소사용 후에는 허가된 표면 소독제를 거즈나 부드러운 천에 적셔 표면과 렌즈를 닦으십시오. 허가되지 않은 세척제는 광중합기를 손상시킬 수 있습니다.
허용되는 세척제:•		•		

70% 이소프로필 알코올
70% 에탄올라이트 쉴드의 일반적인 청소:표면 소독제를 사용하여 VALO Cordless 라이트 쉴드를 청소하십시오. 오토클레이브하지 마십시오.
수리사용자 수행 유지 보수1)배리어 슬리브를 사용하여 치과용 복합재가 렌즈 표면에 달라붙는 것을 방지하십시오. 필요한 경우 플라스틱 또는 스테인리스 스틸 치과 기구를사용하여 부착된 복합재를 조심스럽게 제거합니다. 렌즈를 손상시킬 수 있는 도구를 사용하지 마십시오.
2)
라이트 미터는 크게 다르며 특정 라이트 가이드 팁과 렌즈를 위해 설계되었습니다. Ultradent는 표준 전원 모드에서 정기적으로 출력을 확인할 것을254경화라이트 보관 및 운송:• 온도: +10°C ~ +40°C (+50°F ~ +104°F)• 상대 습도: 10 % ~ 95 %• 주변 압력: 500 hPa 내지 1060 hPa전자 폐기물 (경화라이트, 충전기, 배터리 및 전원공급 장치 등)을 처분할 때는 지역 폐기물 및 재활용 지침을 따르십시오.

기술적 고려사항
부속품

항목

CE 정보

VALO 코드리스 라이트 실드

255

항목
VALO 충전기

CE 정보기술 정보 / 데이터효과적인 복합재 경화 파장 대역:효과적인 복합 경화 파장 대역(활용 가능한 파장 범위: 385-515nm)

유효 하
파장 대역
395-415 nm의유효 상위 파장 대역
440-480 nm의

스펙트럼 복사력(mW/nm)

VALO 배터리

Xtra

고출력

표준

WAVELENGTH (nm)

속성
렌즈

파장 범위

정보 / 사양

지름 11.7 mm• 활용 가능한 파장 범위: 385 - 515nm• 최대 파장: 395 - 415nm 및 440 - 480nm

공칭 복사열 발산도 비교 차트측정 기구라이트 세기표

측정장치의조리개 구멍
표준 전력(±10%)

고전력 플러스
(±10%)

엑스트라 전력
(±10%)

256‡ 기가헤르츠(Gigahertz) 스펙트럼 분석장치
발산도

총 전력

900 mW/cm²

970 mW

1500 mW/cm²

1615 mW

15mm

2100 mW/cm²

VALO 그랜드경화 라이트

등급: IEC 60601-1 (Safety), IEC 606011-2 (EMC)충전기 전원공급
장치

Output - 12VDC at 500mA
Input - 100VAC to 240VAC
Ultradent P/N 5930 VALO Charger
Power Supply with international pluginserts방사 발산도는 기기 기능, 측정 방법 및조명 배치에 따라 달라집니다.
‡ 방사 발산도는 Gigahertz-Optik스펙트럼 분석기로 측정했을 때 ISO
10650을 준수합니다.

15mm

2260 mW중량:• 배터리 포함 : 6 온스 (190g)• 배터리 제외: 5 온스 (150g)
VALO 충전기 전원공급 장치는 의료용 등급 클래스 II 전원공급장치이며, MAINS 전원으로부터 격리됩니다

Ratings: IEC 60601-1 (Safety)
Cord Length - 6 feet (1.8 meters)
VALO Charger Power Supply is a Medical Grade Class II power
supply and provides isolation from MAINS power

257

속성

정보 / 사양

VALO 충전기

VALO 배터리작동 조건충격 계수(Duty
Cycle)

문제의 진단 및 해결전자기 방출에 관한 지침 및 제조자의 선언

VALO 3.6VDC 리튬철인산염 스마트 배터리 충전기:• 완전히 충전되면 자동으로 꺼짐• 결함있는 배터리 자동 감지• 보호: 열, 과충전, 합선, 역 극성⸰ 적색 LED – 충전 중⸰ 녹색 LED – 비어 있거나 완전히 충전됨⸰ LED 꺼짐 - 합선 t• 충전 시간: 1 - 3시간
등급: CE, WEEE

온도: +10°C ~ +32°C (+50°F ~ +90°F)상대 습도: 10 % ~ 95 %주변 압력: 700 hPa 내지 1060 hPa경화라이트는 단시간 작동용으로 설계되었습니다. 최대 주변온도 (32°C)에서 1분 연속 순환시 30분 동안 꺼짐 (냉각시간).

아래에 제시된 해결책으로 문제가 해결되지 않으면, Ultradent에 800.552.5512로 연락하십시오. 미국 이외의 지역에서는 Ultradent 대리점 또는 치과딜러에게 문의하십시오.

라이트가 켜지지 않음라이트가 원하는 시간 동안 켜있지 않음라이트가 레진을 제대로 경화하지않음배터리가 충전되지 않음충전기가 배터리를 충전하지 않음모드 또는 시간 간격을 변경할
수 없음

기타 정보

순응

전자기 환경 – 안내무선주파수 방출CISPR 11클래스 B경화 라이트는 내부 기능을 위해서만 전기 및 전자기 에너지를 사용합니다. 따라서,무선주파수(RF) 방출은 매우 낮으며 근처의 전자 장비에 간섭을 일으키지 않습니다.

전압변동 / 깜빡거림 방출
IEC 61000-3-3

해당 없음

RF 방출
CISPR 11

충전식: 안전한 화학 리튬 철 인산염(LiFePO4) RCR123A• 작동 전압: 3.2VDC, 400mAH (와트시 필요)
등급: UL 62133-2, CE, RoHS, WEEE셀은 용량의 70%에 도달하기 전까지 2000회 이상의 충전/방전 주기를 견딜 수 있습니다. 배터리는 3-5년마다교체하는 것이 좋지만 수명은 사용에 따라 다를 수 있습니다. 필요에 따라 교체하세요.

문제방출 테스트가능한 해결책

1) 절전 모드에서 깨우려면 시간/모드 또는 전원 단추를 누르십시오.
2) 빨간색 배터리 부족 표시등을 보고 배터리 충전 상태를 확인하십시오.
3) 새 배터리가 장치에 바르게 삽입되었는지 확인합니다.
4) 빨간색과 노란색 경고 LED가 깜박이면 이는 경화 라이트가 내부 온도 안전 한계에 도달했음을 의미합니다.
경화 라이트를 10분 동안 식히거나 시원하고 촉촉한 수건으로 기기를 빨리 식히십시오.
5) 빨간색 경고 LED가 깜박이고 경고음이 계속 울리면 Ultradent 고객 서비스 센터에 수리를 의뢰하십시오.
1) 모드 및 시간 표시등에서 정확한 시간 입력을 확인합니다.
2) 배터리 부족 표시등을 보고 배터리 충전 상태를 확인하십시오.
3) 새 배터리가 장치에 올바르게 삽입되었는지 확인하십시오.

1) 렌즈에 경화된 레진이나 복합재 잔여물이 있는지 확인합니다.
2) 적절한 호박색 자외선 눈 보호 기능을 사용하여 LED 라이트가 작동하는지 확인하십시오.
3) 노출계로 전력 레벨을 점검합니다. 노출계를 사용하는 경우, Ultradent는 경화 라이트를 표준전력 모드에서점검 할 것을 권장합니다.
주: 일반적인 노출계의 부정확성과 경화 라이트에서 사용되는 사용자 정의 LED 팩의 부정확성으로 인해 실제수치 출력은 왜곡되어 나타납니다. 노출계(light meter)마다 크게 다르며, 특정 라이트 가이드 팁과 렌즈에 맞게설계되어 있습니다.
4) 경화 수지의 유효 기간을 확인합니다.
5) 제조자의 권장사항에 따라 적절한 기술을 따르고 있는지 확인하십시오.

1) 배터리가 올바른 방향으로 충전기에 삽입되었는지 확인하고, 배터리를 1-3시간 동안은 충전되도록 하십시오.
2) 충전기의 적색 표시등이 녹색으로 바뀌지 않으면 Ultradent 고객 서비스부에 연락하여 교체용 배터리나충전기를 주문하십시오.
3) 충전기의 녹색이나 적색 표시등이 켜지지 않으면, Ultradent 고객 서비스부에 연락하여 충전기나 AC어댑터를 주문하거나 교체하십시오.
1) 충전기와 AC 어댑터가 작동되는 전원 콘센트에 연결되어 있는지 확인하십시오.
2) 충전기의 녹색이나 적색 표시등이 켜지지 않으면, Ultradent 고객 서비스부에 연락하여 새 충전기나 AC어댑터를 받으세요.

1) 일련의 경고음으로써 경화 라이트가 잠금 해제되었음을 나타낼 때까지, '시간/모드' 버튼과 '전원' 버튼을 모두누른 채 기다립니다.

고조파 방출
IEC 61000-3-2

그룹 1

해당 없음

경화 라이트는 배터리를 사용하며, EMI, RF 또는 서지 억제의 영향을 받지 않습니다.

배터리로 전원이 공급되므로, 고조파 방출 및 전압 변동 테스트는 경화 라이트에 적용되지않습니다.
경화 라이트는 가정용 건물에 공급되는 공공 저전압 전원공급 네트워크에 직접 연결된가정 시설물 등 모든 시설에서 사용하기에 적합합니다.

전자기 내성에 대한 지침 및 제조자의 선언경화 라이트는 아래 명시된 전자파 환경에서 사용하도록 설계되었습니다. 고객 또는 사용자는 그러한 환경에서 사용되는지 확인해야 합니다.
내성 시험

IEC 60601 테스트 수준규정준수 수준전자기 환경 가이드전기적으로 빠른 순간전류(transients)/파열
IEC 61000-4-4전원 공급 라인의 경우 ±2kV
입력/출력 라인의 경우 ± 1kV

전원 공급 라인의 경우 ±2kV
주 1: 경화 라이트에는 I/O포트가 없습니다.

경화 라이트는 배터리로 작동하며, AC MAINS전원에 연결할 수 없습니다.

전원공급 장치 입력 라인의전압, 순간 전압, 단락, 정전및 변동<5 % U(0.5 사이클 동안 U에서 95%
이상 감소)

정전기 방전(ESD)
IEC 61000-4-2± 8 kV 접촉± 15kV 공기

서지(Surge)
IEC 61000-4-5± 1 kV 라인에서 라인± 2 kV 라인에서 접지

IEC 61000-4-11

40% U(5 사이클 동안 U에서 60%
감소)

70% U
(25 사이클 동안 U에서 30%
감소)<5 % U(5초 동안 U에서 95% 이상감소)전력 주파수
(50/60 Hz) 자기장
IEC 61000-4-8

30 A/m주 1: 경화 라이트에 포트 또는 접근 가능한 I/O 라인이 없습니다.

± 8 kV 접촉± 15kV 공기± 1 kV 라인에서 라인± 2 kV 라인에서 접지<5 % U(0.5 사이클 동안 U에서 95%
이상 감소)
40% U(5 사이클 동안 U에서 60%
감소)

70% U
(25 사이클 동안 U에서 30%
감소)<5 % U(5초 동안 U에서 95% 이상감소)주 2: 자체 회복
30 A/m물리적 환경은 다음으로 제한되어야 합니다:1) IP 코드: IP202) 액체에 담그지 마십시오.
3) 가연성 가스 주변에서는 사용하지 마십시오.
단위는 Non-APG 및 Non-AP입니다.
4) 보관 습도 범위: 10% - 95%5) 보관 온도 범위: 10° C - 40° C경화 라이트는 배터리로 전원이 공급되기 때문에,전기 과도 현상, 서지, 전압 강하, 단락, 중단 또는
AC MAINS 전력 변동의 영향을 받지 않습니다.
액세서리 배터리 충전기는 위의 적용을 받을 수있지만, VALO 코드리스 경화 라이트와 분리되어,VALO 코드리스 경화 라이트의 작동에 큰 영향을미치지 않습니다.
경화 라이트의 배터리 전압이 4VDC로 떨어지면,기기가 작동하지 않습니다. VALO 코드리스경화 라이트가 꺼지게 됩니다. 새 배터리를 넣고적절한 전원 수준으로 복원되면, 경화 라이트가다시 시작되고 전원이 꺼지기 전에 동일한상태로 돌아갑니다. 경화 라이트는 정전 시 자체복구됩니다.

전원 주파수 자기장은 전형적인 주거용, 가정건강관리용, 상업용, 병원용 또는 군사용환경에서 일반적인 위치의 특성을 가진 높이에있어야 합니다.

전자기 방출에 관한 지침 및 제조자의 선언경화 라이트는 아래 명시된 전자파 환경에서 사용하도록 설계되었습니다. 고객 또는 사용자는 그러한 환경에서 사용되는지 확인해야 합니다.
258

259

VALO™ 大款비 수명 지원 시스템에 대한 전자기 내성에 관한 지침 및 제조자의 선언경화 라이트는 아래 명시된 전자파 환경에서 사용하도록 설계되었습니다. 고객 또는 사용자는 그러한 환경에서 사용되는지 확인해야 합니다.
IEC 60601 테스트
수준

내성 시험

규정준수 수준전도성 무선주파수

3 Vrms3 VrmsIEC 61000-4-6

150kHz ~ 80MHz

150kHz ~ 80MHz

방사되는
무선주파수

3V/m

IEC 61000-4-3

80 MHz ~ 2.5 GHz

3V/m

80 MHz ~ 2.5 GHz전자기 환경 가이드휴대형 및 이동형 RF 통신 장비는 케이블을 포함하여 경화 라이트의 어느 부분에대해서도 송신기의 주파수에 적용되는 방정식에서 계산된 권장 이격 거리보다 더가깝게 사용해서는 안 됩니다.
권장 이격 거리:80 MHz 내지 800 MHz

0.01

100

80 MHz – 800 MHz

0.35m

11.7m

3.5m

3.7m

1

송신기의 주파수에 따른 분리 거리
(미터)

1.7m

10

800 MHz – 2.5 GHz

0.07m

0.11m

0.22m

1.11m

2.22m

0.7m

7.0m

경화 라이트는 IEC 60601-1-2:2014에 따라 테스트되었으며 80MHz 내지 2.5GHz 사이에서 10 V/m의 복사 전계 강도로 통과되었습니다. 위의공식에서 3 Vrms의 값은 V1에 해당하고 10 V/m의 값은 E1에 해당합니다.
위에서 언급하지 않은 최대 출력으로 정격된 송신기의 경우, 권장 분리 거리 d (m)는 송신기의 주파수에 적용할 수 있는 방정식을 사용하여 추정할 수
있습니다. 여기서 P는 송신기 제조자에 따를 때, 송신기의 최대 출력 전력 정격(와트)입니다.
주 1 80 MHz와 800 MHz에서, 높은 주파수 범위에 대한 이격 거리가 적용됩니다.
주 2 이 지침이 모든 상황에서 적용되는 것은 아닙니다. 전자파 전파는 구조물, 물체 및 사람의 흡수 및 반사에 영향을 받습니다.
260

1 – VALO Grand 光固化机
4 – Ultradent VALO 可充电电池
1 – Ultradent VALO 电池充电器，带医疗级 12VDC 交流电源适配器
1 – VALO 隔离套样品包1 – VALO 大遮光罩
1 – 带双面胶带的光固化表面安装支架

照片

按钮和指示灯电源按钮（顶部或底部）

图例

定时指示灯

电池电量低 – 指示灯

电池电量低 – 指示灯다음 기호가 표시된 장비 부근에서 간섭이 발생할 수 있습니다:0.035m

0.37m

•
•
•
•
•
•

전자기장 조사에 의해 결정된 고정 RF 송신기의 전계 강도는 각 주파수 범위에서의준수 수준보다 낮아야 합니다.b

0.12m

0.1

VALO 大款产品组件:P는 송신기 제조업체에 따른 송신기의 최대 출력 정격(W)이며, d는 권장 분리 거리(m)입니다.

경화 라이트는 복사된 RF 장해가 제어되는 전자파 환경에서 사용하기 위한 것입니다. 경화 라이트 사용자는 통신 장비의 최대 출력에 따라, 아래에서권장하는 바와 같이, 휴대형 및 이동형 RF 통신 장비(송신기)와 경화 라이트 사이의 최소 거리를 유지함으로써 전자기 간섭을 방지할 수 있습니다.
150 kHz – 80 MHz凭借其宽带光谱，VALO Grand 固化灯旨在聚合波长范围为 385-515nm 的所有光固化产品（符合 ISO 10650）。
VALO Grand 固化灯使用 Ultradent VALO 可充
电电池和电池充电器。
固化灯设计用于放置在标准牙科综合治疗机支架中，或者可以使用套件附带的支架进行定制安装。

800 MHz 내지 2.5 GHz휴대형 및 이동형 RF 통신 장비와 VALO 코드리스 경화 라이트 사이의 권장 이격 거리에 관한 지침 및 제조자의 선언송신기의 정격최대 출력(와트 단위의 전력)

产品描述

控件概述:

주 1 80 MHz와 800 MHz에서, 더 높은 주파수 범위가 적용됩니다.
주 2 이 지침이 모든 상황에서 적용되는 것은 아닙니다. 전자파 전파는 구조물, 물체 및 사람의 흡수 및 반사에 영향을 받습니다.
a라디오 (셀룰러 / 무선) 전화기 및 육상 이동 라디오, 아마추어 무선, AM 및 FM 라디오 방송 및 TV 방송을 위한 기지국과 같은 고정 송신기의 자장강도는 이론적으로 정확하게 예측할 수 없습니다. 고정 RF 송신기로 인한 전자기 환경을 평가하려면 전자기 현장 조사가 고려되어야 합니다. 경화라이트가 사용된 위치에서 측정된 전계 강도가 위의 해당 RF 준수 레벨을 초과하면, 정상 작동을 확인하기 위해 VALO 코드리스 경화 라이트를관찰해야 합니다. 비정상적인 성능이 관찰되면 경화 라이트 방향을 바꾸거나 재배치하는 등의 추가 조치가 필요할 수 있습니다.
b
150 kHz ~ 80 MHz의 주파수 범위에서 전계 강도는 3 V/m 미만이어야 합니다.

ZH

固化灯

定时/模式更改按钮

(底部)

(顶部)

对于所描述的产品，使用前请仔细阅读并理解所有说明和 SDS 信息。

适用症/预期目的

固化光活化牙科修复材料和粘合剂的光源。

警告和注意事项风险组 2请注意，本产品发射紫外线。
暴露可能会导致眼睛或皮肤刺激。
进行适当的遮蔽。
请注意，本产品可能会发射危险光射线。
请勿直视手术灯。
可能对眼睛有害。

• 请勿直视光输出。
使用固化灯时，患者、临床医生和助手应始终佩戴琥珀色紫外线护目镜。
• 为防止触电风险，不允许改装本设备。
仅使用随附的 Ultradent VALO 电源和插头适配器。
如果这些组件损坏，请勿使用并致电 Ultradent 客户服务部门订购更换件。
• 如果使用距离小于 30 厘米（12 英寸），便携式射频通信设备可能会降低性能。
• 仅使用授权的配件、电缆和电源，以防止操作不当、电磁辐射增加或电磁抗扰度降低（请参阅电磁辐射部分）。
• 为避免处理电池时发生电气火灾的风险：⸰ 请勿将充电电池与不可充电电池或其他类型的电池混合使用。
⸰ 请勿尝试为不可充电电池充电。
对不可充电电池充电可能会导致人身伤害或财产损失。
⸰ 请勿用任何类型的液体对电池、电池触点、充电器或交流电源适配器进行高压灭菌或喷射。
如果电池充电器的触点出现腐蚀，请致电 Ultradent 客户服务部门订购更换件。
⸰ 请勿在易燃材料周围为电池充电。
⸰ 请勿将充电器放在临床手术室中。
• 为避免受伤风险，请勿使用已腐蚀（生锈）、凹陷、散发气味或液体、包装破损或丢失或有其他损坏的电池。
请致电 Ultradent 客户服务部订购更换电池。
替代电池可能会导致故障。
• 仅使用推荐的电池。
• 为了防止热刺激或受伤的风险，请避免连续固化循环，并且在任何模式下都不要将口腔软组织近距离暴露超过 10 秒。
如果需要更长的固化时间，请使用多个
固化周期并在周期之间休息一段时间，或使用双重固化产品以避免加热软组织。
• 治疗患有不良光生物学反应或过敏的患者、正在接受化疗的患者或正在接受光敏药物治疗的患者时要小心。
• 该装置可能容易受到强磁场或静电场的影响，这可能会破坏编程。
如果您怀疑发生了这种情况，请立即拔下设备电源插头，然后将其重新插入电源插座。
不遵循随附的处理说明可能会导致固• 请勿使用腐蚀性或研磨性清洁剂、高压灭菌器擦拭固化灯，或浸入任何类型的超声波浴、消毒剂、清洁溶液或液体中。
化灯无法运行。
• 为避免损坏设备，请勿将手指、仪器或其他物体插入光固化机的电池盒中。
如果有疑问，请致电 Ultradent 客户服务部。
• 为避免损坏设备，请勿尝试清洁金色触点或电池盒的任何部分。
• 为了防止交叉污染的风险，隔离套仅供单个患者使用。
• 为了降低腐蚀风险，请在使用后取下隔离套。
• 为了降低树脂固化不足的风险，如果透镜损坏，请勿使用固化灯。
261

分步说明
准备

1) 使用固化灯前请为电池充电（请参见电池维护章节）。
2) 将固化灯放入标准牙科综合治疗台安装支架或安装支架附件上，直到做好使用准备。
3) 在每次使用之前，在固化灯上放置一个新的屏障套。

快速模式指南:电源按钮安装卫生屏障套:注:

VALO 无绳遮光罩:定时选项

VALO 无绳遮光罩为椭圆形，可旋转以实现最大化的使用，而且可以与透明屏障套配套使用。

要更改定时

使用

定时间隔：5、10、15、和 20 秒。
• 固化灯初次通电时默认这一模式。
模式/状态指示灯将为绿色，四个绿色定时指示灯将亮起，表示标准功率模式。
• 要更改定时间隔，请快速按定时/模式按钮。
• 按电源按钮可进行固化。
要在完成定时间隔之前停止固化，请再次按电源按钮。

要更改模式
图例

快速固化指南:固化模式：外加功率模式

定时间隔：仅 3 秒（注：外加功率模式在每次固化循环结束时都有一段 2 秒的安全延时，以限制在连续固化期间进行加热。
延时结束时，笛音将响起，表示装置可以继续使用）。
• 在标准功率模式下，按下并保持住定时/模式更改按钮 2 秒，释放，然后再按下并保持住 2 秒，再释放。
模式/状态指示灯将为橙色并闪烁，并且其中三个绿色
定时指示灯将亮起并闪烁，表示外加功率模式。
• 按电源按钮可进行固化。
要在完成定时间隔之前停止固化，请再次按电源按钮。
• 要返回标准功率模式，请按下并保持住定时/模式按钮 2 秒，然后释放。
模式/状态指示灯将为绿色，绿色定时指示灯将亮起，表示标准功率模式。

固化模式
标准

高功率+
超功率

快速警告指南:功率等级警告更换电池• 电池电量低：慢闪
• 关闭：3 声笛音，闪烁
• 禁止操作睡眠模式：不活动 60 秒后，固化灯将进入睡眠模式，由模式/状态指示灯的慢速闪烁指示。
拾取或触摸装置将唤醒固化灯并自动将其返回到上次使用的设置。
为了最大限度地延长电池寿命，请在不使用时保持固化灯不受干扰。

5秒

10秒

15秒

20秒

1秒

快速按下然后释放定时按钮可切换定时选项。

2秒

3秒

4秒

仅3秒

按下并保持住定时按钮 2 秒，然后释放。
固化灯将切换到下一个模式。
稳亮 LED（焦耳）

固化模式：高功率加模式

定时间隔：1、2、3、和 4 秒。
• 在标准功率模式下，按下并保持住定时/模式更改按钮 2 秒，然后释放。
模式/状态指示灯将为橙色，四个绿色定时指示灯将亮起并闪烁，表示高功率模式。
• 要更改定时间隔，请快速按定时/模式按钮。
• 按任何一个电源按钮可进行固化。
要在完成定时间隔之前停止固化，请再次按任何一个电源按钮。
• 要返回标准功率模式，请按下并保持住定时/模式更改按钮 2 秒，然后释放，这将切换至外加功率模式。
再次按下并保持住 2 秒，然后释放。
模式/状态指示灯
将为绿色，四个绿色定时指示灯将亮起，表示标准功率模式。

外加功率3200 兆瓦/厘米2

定时按钮• 使用卫生屏障套将减少光输出 5-10%。
由于固化灯的高输出功率，固化效果已被证明是基本相当的。
• 固化灯必须在每位患者使用后使用适当的清洁和/或消毒剂进行清洁和消毒。
请参见标题为处理的章节。

操作
固化模式：标准功率模式

高功率加1600 兆瓦/厘米2

模式/定时 LED卫生屏障套定制安装在固化灯上，保持固化灯的表面清洁。
屏障套有助于防止交叉污染，有助于保持牙科复合材料不粘附到镜头和固化灯的表面上，并且预防因清洁溶液而变色和腐蚀。

1) 每种功率模式用于使用光引发剂固化牙科材料。
请参见快速模式指南了解建议固化时间。
注：固化灯经编程将依次从标准功率到高功率加再到外加功率模式进行切换。
比如，要从标准功率模式切换到外加功率模式，必须先切换到高功率加模式，然后再切换至外加功率模式。
2) 固化灯存储最近使用的定时间隔和模式，每当更改模式或拆除电池时，它均将默认返回该模式。

标准功率1000 兆瓦/厘米2

模式

闪烁 LED

功率* (mW)
970

辐照度* (mW/cm2)

总曝光时间（秒）900

5/10/15/20

900

970

1,620

1,500

2,260

2,100

能量**

5/10/15/20

4.8J/9.7J/14.5J/19.4J

1/2/3/4

1.6J/3.2J/4.8J/6.5J

4.8J/9.7J/14.5J/19.4J

3

6.8J

*렌즈 끝에서 복합재의 상단 표면까지 2mm 거리에서의 공칭 출력.
**사이클당 총 에너지(줄) 섹션의 값은 가장 가까운 10분의 1로 반올림됩니다.
温度警告
允许冷却

• 3 声笛音• 慢闪• 禁止操作服务警告致电客户服务部进行维修• 无声音• 闪烁，每 2 秒一次• 允许操作

LED 警告致电客户服务部进行维修• 连续 3 声笛音• 单快闪• 禁止操作清理1) 每次患者使用之后，将使用过的屏障套弃置在常规废弃物中。
2) 参见处理章节。

安装支架说明1) 支架应安装在平整、无油的表面上。
2) 用医用酒精清洁表面。
3) 撕掉支架的胶带。
4) 定位支架，以使在移除时固化灯向上提升。
牢固地按压到位。

维护

光固化机的一般清洁每次使用后，用经认可的表面消毒剂润湿纱布或软布，然后擦拭表面和镜片。
未经授权的清洁剂可能会损坏光固化机。

可接受的清洁剂：•70% 异丙醇
•70% 乙醇

262

遮光罩的一般清洁：使用任何表面消毒剂清洁 VALO Cordless 遮光罩。
请勿高压灭菌。
修复
用户执行的维护
1) 使用隔离套管防止牙科复合材料粘附到镜片表面。
如有必要，使用塑料或不锈钢牙科器械小心地去除任何粘附的复合材料。
请勿使用会损坏镜头的工具。
2) 测光表差异很大，专为特定的光导尖端和镜头而设计。
Ultradent 建议定期检查标准功率模式下的输出。
注意：由于普通照度计和固化灯中的定制 LED 组的不
263

准确，真实的数字输出将会出现偏差电池维护充电和更换电池
固化灯配有 4 节可充电磷酸铁锂电池。
Ultradent 建议仅使用 VALO 充电电池。

产品
VALO 无绳遮光罩

如何给电池充电：1) 将充电器和电池放置在不会接触液体或消毒喷雾的区域。
2) 将电池插入充电器，正极 (+) 端指向充电器上的指示灯。
3) 绿灯表示电池已准备好使用。
4) 电池充电需要 1-3 小时。
将电池放在充电器中直至准备使用。
注意：如果电池充电时间超过三小时后充电器上的红灯没有变成绿色，则电池可能会降级（损坏）并且无法充电。
尝试使用新电池或致电 Ultradent 客户服务部
门订购一套新的充电电池。

电池电量不足：当电池电量不足指示灯呈红色闪烁时，固化灯会向用户发出更换电池的信号。
如果电池电量过低，将会发出 3 声蜂鸣警告声，并且固化灯将不允许进一步操作，直到电池充电或插入新电池。
（参见快速警告指南）仅限 VALO Cordless/VALO Grand（仅限 UD）
充电时间和电池寿命：固化灯充满电后的电池寿命取决于模式/时间间隔、电池类型、使用量和 LED 效率。
•建议充电间隔：根据需要或在低电量指示灯亮起时充电。
预期寿命取决于使用情况。
•额外电池：固化灯配有 4 节可充电电池。
仅限 VALO Ortho Cordless（仅限 UD）
充电时间和电池寿命：固化灯充满电后的电池寿命取决于模式/时间间隔、电池类型、使用量和 LED 效率。
一般来说，充电电池应可持续大约 400 个固化周期。
•建议充电间隔：当低电量指示灯亮起时，或大约每 1 – 2 周充电一次，具体取决于使用情况。
•额外电池：固化灯配有 4 节可充电电池。
•电池预期寿命：可充电磷酸铁锂电池可充电约 1000 至 2000 次。
根据正常使用和适当保养，电池可持续使用长达五年，但应根据需要进行更换。
厂家维修1) 维修只能由授权维修人员进行。
Ultradent 为服务人员提供执行维修的文档。
2) 将设备送去维修、保养或校准时，请务必从固化灯和充电器中取出电池。
将电池、充电器、适配器和光固化机分别包装在回收箱中。
3) 按照当地法规运输电池。
保修
Ultradent Products, Inc.（“Ultradent”）保证，本产品自购买之日起 5 年内，按照产品附带的操作说明进行操作时，(i) 在所有重大方面均符合产品附带的
Ultradent 文档中规定的规格；(ii) 材料和工艺上不存在缺陷。

VALO 充电器
VALO 电池

技术信息/数据

有效的复合材料固化波段:

有效的复合材料固化波段：
有效复合材料固化波长带（可用波长范围：385-515 nm）
有效下限波长带
395-415 纳米

此有限保修不可转让，仅适用于原始购买者，不适用于产品的后续所有者。
本有限保修不涵盖任何其他配件，例如但不限于电池、充电器、适配器或自适应镜头。
如果产品因疏忽、滥用、误用、事故、修改、篡改、改动或未能遵循适用的使用说明而出现故障或损坏，则本有限保修无效。
仅出于示例目的，跌落和损坏的产品不在本保修范围内。
为了符合本有限保修的资格，购买证明（例如销售收据或类似文件）必须与有缺陷的产品一起提交给 Ultradent。

该产品旨在与隔离套一起使用。
请参阅上文了解维护和一般清洁。
储存和处置

如果储存固化灯 2 周以上或将其打包运送，请务必取出电池。
如果电池长时间留在装置中不进行充电，其可能会变得不能工作或无法充电。
请勿将电池储存在温度超过 40°C (104°F) 或受阳光直射的地方。

技术考虑因素
附件

属性
镜头

波长范围

信息/规格

直径 11.7 毫米• 可用波长范围：385 – 515 纳米• 峰值波长：395 – 415 纳米和 440 – 480 纳米

Nominal Radiant Exitance Comparison Chart

测量仪器
光强度表

仪表孔径
标准功率
(±10%)
高功率加(±10%)
外加功率(±10%)

264

高功率

瓦维尔英格（纳米）固化灯储存和运输:• 温度：+10°C 至 +40°C（+50°F 至 +104°F）• 相对湿度：10% 至 95%• 环境压力：500 百帕至 1060 百帕处置电子废弃物（即固化灯、充电器、电池和电源系统）时，请遵守当地废弃物及回收准则。

超功率标准符合此处规定的保修条件的有缺陷的产品将由 Ultradent 自行决定进行维修或更换。
在任何情况下，Ultradent 对产品的责任均不超过购买者支付的购买价格。
在任何情况下，Ultradent 均不对因使用本产品而引起的或与之相关的任何间接、偶然、可预见、不可预见、特殊或后果性损害承担责任。

处理有效上波长带
440-480 纳米光谱辐射功率 (毫瓦/纳米)如何更换/插入电池：1) 逆时针旋转四分之一圈，取下后盖。
2) 取出电池。
3) 首先插入新电池的正极 (+) 侧。
4) 对齐并轻轻推动，同时顺时针旋转，重新装上后盖。
完全连接后，盖子会发出咔哒声。
5) 该装置已准备就绪，可供使用。

CE 信息‡ Gigahertz spectrum analyzer

Exitance

Total Power

900 兆瓦/厘米²970 兆瓦1500 兆瓦/厘米²1615 兆瓦2100 兆瓦/厘米²2260 兆瓦15 毫米

辐射出射度将根据仪器功能、测量方法
和光位置而变化。
‡ 使用 Gigahertz-Optik 频谱分析仪测量时，辐射出射度符合 ISO 10650。

15 毫米

265

属性

信息/规格

VALO 大款固化灯

评级：IEC 60601-1（安全）、IEC 60601-1-2（电磁兼容性）充电器电源系统输出 - 500 毫安时为直流 12 伏
输入 - 交流 100 伏至交流 240 伏
Ultradent P/N 5930 VALO 充电器电源系
统，带国际通用插头VALO 充电器

VALO 电池
操作条件

故障排除

工作循环

重量:
• 含电池：
6 盎司（190 克）• 不含电池：5 盎司（150 克）
尺寸：（8 x 1.28 x 1.06）英寸（203 x 32.5 x 27）毫米

评级：IEC 60601-1（安全）电源线长 - 6 英尺（1.8 米）
VALO 充电器电源系统为医用级 II 类电源系统，提供与电网电
源的隔离。

VALO 直流 3.6 伏锂铁磷酸盐智能电池充电器:• 充满电后自动关闭• 自动检测有缺陷的电池• 保护：热、过充、短路、反向极性⸰ 红色 LED – 充电⸰ 绿色 LED – 空电量或充满电⸰ LED 熄灭 – 短路
• 充电时间：1 – 3 小时
等级：CE、WEEE

可充电：安全化学磷酸铁锂 (LiFePO4) RCR123A•工作电压：3.2VDC，400mAH（需要瓦时）
评级：UL 62133-2、CE、RoHS、WEEE
电池在达到其容量的 70% 之前，额定充电/放电循环次数超过 2000 次。
建议每 3-5 年更换一次电池，但使用寿命可能会因使用情况而异。
根据需要更换。
温度：+10°C 至 +32°C（+50°F 至 +90°F）
相对湿度：10% 至 95%
环境压力：700 百帕至 1060 百帕

固化灯设计用于短时操作。
在最高环境温度 (32°C) 下，背对背循环打开 1 分钟，关闭 30 分钟（冷却期）。

如果以下所建议的方法未能排除故障，请致电Ultradent，电话：800.552.5512。
在美国境外，请致电您的Ultradent经销商或牙科经销商。
问题

灯不亮起

灯不保持亮起达到所需的时间灯不正常固化树脂电池不充电充电器不给电池充电无法更改模式或定时间隔

其他信息
266

可能解决方案

1) 按定时/模式或电源按钮，以从节电模式唤醒。
2) 检查红色电量低指示灯，以了解电池电量状态。
3) 检查新电池是否正确插入本装置。
4) 如果红色和黄色警告 LED 闪烁，则表示固化灯已达到其内部温度安全极限。
让固化灯冷却 10 分钟，或使用凉爽的湿毛巾快速冷却装置。
5) 如果红色警告 LED 持续闪烁并发出笛音，请致电Ultradent客户服务部以进行维修。

关于电磁辐射的指导意见和制造商声明

固化灯旨在在以下指定的电磁环境中使用。
客户或用户应确保其在这样的环境中使用。
辐射测试

合规

电磁环境 - 指导意见射频发射CISPR 11B类

固化灯仅为其内部功能使用电和电磁能。
因此，其任何射频辐射均非常低，不可能对
附近的电子设备造成干扰。

电压波动/闪烁辐射 IEC
61000-3-不适用射频辐射
CISPR 11
谐波辐射IEC 61000-3-21) 确保电池以正确的方向插入充电器，并允许电池充电 1-3 小时。
2) 如果充电器上的红色指示灯不变为绿色，请致电Ultradent客户服务部，以订购更换电池和/或充电器。
3) 如果充电器上绿色指示灯或红色指示灯均不亮起，请致电Ultradent客户服务部，以订购或更换充电器
和/或交流适配器。
1) 确保充电器已插入，且交流适配器已插入工作电源插座。
2) 如果充电器上的绿色或红色指示灯不亮起，请致电Ultradent客户服务部，以订购新充电器和/或交流适配器。
1) 同时按住时间/模式和电源按钮，直到发出一连串笛音，表示固化灯已解锁。

固化灯使用电池，不受电磁干扰、射频或浪涌抑制的影响。

谐波辐射和电压波动测试不适用于固化灯，因为它由电池供电。

不适用

固化灯适用于所有建筑设施，包括住宅和直接连接到为住宅提供生活用电的公共低压供电网络的建筑设施。

关于电磁抗扰性的指导意见和制造商声明

固化灯旨在在以下指定的电磁环境中使用。
客户或用户应确保其在这样的环境中使用。
抗扰性测试

静电放电 (ESD)电快速瞬变/突变
IEC 61000-4-4
浪涌
IEC 61000-4-5

电压、骤降、短路、中断和电源系统输入线路上的变化
IEC 61000-4-11

IEC 60601 测试级别合规水平

± 2 千伏（对于供电线路）± 1 千伏（对于输入/输出线路）± 2 千伏（对于供电线路）
固化灯由电池供电，不能连接到交流电网
注 1：固化灯无输入/输出端口 电源上。

<5% U
（0.5 个循环> 95% 的骤降，单位为 U）<5% U（0.5 个循环> 95% 的骤降，单位为 U）± 8 千伏（接触）± 15 千伏（空气）IEC 61000-4-21) 检查模式和定时指示灯的时间输入是否正确。
2) 检查电量低指示灯，以了解电池电量状态。
3) 检查新电池是否正确插入本装置。

1) 检查镜头上是否有残留的固化树脂/复合材料。
2) 戴好适当的琥珀色紫外线护眼罩，确认 LED 指示灯正在工作。
3) 使用测光表检查功率等级。
如果使用测光表，Ultradent建议在标准功率模式下检查固化灯。
注：真实的数字输出会因通用测光表的不准确性和固化灯的定制 LED 套件而存在偏差。
测光表各不相同，需针对特定的光导尖嘴和镜头进行设计。
4) 检查固化树脂的有效期。
5) 确保在按制造商的建议遵守正确的技术。

第1组± 1 千伏（线到线）± 2 千伏（线到地线）± 1 千伏（线到线）± 2 千伏（线到地线）

40% U（5 个循环 60% 的骤降，单位为 U）

40% U（5 个循环 60% 的骤降，单位为 U）

70% U（25 个循环 30% 的骤降，单位为 U）<5% U（5 秒> 95% 的骤降，单
位为 U）
电源频率
（50/60 赫兹）磁场

± 8 千伏（接触）± 15 千伏（空气）30 安/米70% U（25 个循环 30% 的骤降，单位为 U）<5% U（5 秒> 95% 的骤降，单
位为 U）
注 2：自动恢复
30 安/米IEC 61000-4-8

电磁环境指导意见

物理环境应局限于以下规定:1) IP 代码：IP202) 请勿浸入液体。
3) 请勿在易燃气体周围使用。
装置不含烷基糖苷和烷基酚。
4) 储存湿度范围：10% - 95%5) 储存温度范围：10° C - 40° C
因为固化灯由电池供电，因此不受电瞬变、
浪涌、电压骤降、短路、中断或交流电网电源变化的影响。

附件电池充电器可能会受上述各项的影响，但它与 VALO 无绳固化灯的操作是分开的，而且对于它的操作并不重要。

如果固化灯的电池电压降至直流 4 伏，则本装
置将不允许操作。
VALO 无绳固化灯将关闭。
当放入新电池并恢复适当的功率等级时，固化灯
将重新启动并恢复到与断电前相同的状态。
在断电的情况下，固化灯将自动恢复。

电源频率磁场应处于典型住宅、家庭医疗保
健、商业、医院或军事环境中典型位置特征
的等级上。

注 1：固化灯未配备任何端口或任何可接入的输入/输出线路。
关于非生命支持系统电磁抗扰性的指导意见和制造商声明

固化灯旨在在以下指定的电磁环境中使用。
客户或用户应确保其在这样的环境中使用。
抗扰性测试

IEC 60601 测
试级别

合规水平

电磁环境指导意见

267

关于非生命支持系统电磁抗扰性的指导意见和制造商声明传导射频

3 伏（均方根）3 伏（均方根）IEC 61000-4-6

150 千赫兹至 80兆赫兹

150 千赫兹至 80
兆赫兹

辐射射频

3 伏/米

3 伏/米

IEC 61000-4-3

使用便携式和移动式射频通信设备时，应与固化灯的任何部分（包括电缆）保
持不少于由适用于发射器频率的方程式所算得的建议间隔距离。
建议间隔距离:80 兆赫兹至 800 兆赫兹
800 兆赫兹至 2.5 千兆赫兹P 为发射机制造商所规定的发射机最大额定输出功率（以瓦 (W) 为单位），d 为
建议间隔距离（以米 (m) 为单位）。

固定射频发射机的场强度，由电磁现场勘测确定a，在每个频率范围内均应低于合规水平b。

80 兆赫兹至 2.5 千 80 兆赫兹至 2.5 千 在标有以下符号的设备的邻近区域都有可能发生干扰:兆赫兹
兆赫兹

注 1 在 80 兆赫兹和 800 兆赫兹下，适用于更高的频率范围。
注 2：这些准则可能不适用于所有情况。
电磁传播受结构、物体和人员的吸收和反射的影响。
a固定发射机的场强度，如无线（蜂窝/无绳）电话和陆地移动无线电、业余无线电、调幅和调频无线电广播和电视广播基站，无法从理论上准确预测。
为了评估由固定射频发射机所造成的电磁环境，应考虑进行电磁现场勘测。
如果在使用固化灯的位置的测得场强度超过上述适用的射频
合规水平，则应观察 VALO 无绳固化灯，以核实其是否正常运行。
如果观察到性能异常，可能需要采取其他措施，比如重新定向或定位固化灯。
b
在 150 千赫兹至 80 兆赫兹频率范围内，场强度应低于 3 伏/米。
关于便携式和移动式射频通信设备与 VALO 无绳固化灯之间建议间隔距离的指导意见和制造商声明

固化灯旨在用于辐射射频干扰受控的电磁环境中。
固化灯的用户可通过根据通信设备最大输出功率维持下文所建议的便携式和移动式射频通信设备（发射机）与固化灯之间的最小距离，可帮助防止电磁干扰。
发射机的额定最大
输出功率（P，以瓦为单位）0.01

80 兆赫兹 – 800 兆赫兹

800 兆赫兹 – 2.5 千兆赫兹

0.12 米0.035 米0.07 米1.7 米0.35 米11.7 米3.5 米

150 千赫兹 – 80 兆赫兹

0.1

0.37 米

10

3.7 米

1

100

按发射机频率的间隔距离（米）

0.11 米1.11 米0.22 米0.7 米2.22 米7.0 米

固化灯已根据 IEC 60601-1-2:2014 进行了测试，并通过了 80 兆赫兹至 2.5 千兆赫兹的 10 伏/米辐射场强度下的测试。
3 伏（均方根）的值对应
于 V1，并且值 10 伏/米对应于上述公式中的 E1。
对于额定最大输出功率未在上表中列出的发射机，建议间隔距离（以米 (m) 为单位）可运用适用于发射机频率的方程式进行估算，此处 P 为发
射机制造商所规定的发射机最大额定输出功率（以瓦 (W) 为单位）。
注 1：在 80 兆赫兹和 800 兆赫兹下，适用于更高频率范围的间隔距离。
注 2：这些准则可能不适用于所有情况。
电磁传播受结构、物体和人员的吸收和反射的影响。

268

269

VALO™ グランド

JA

硬化ライト

製品の説明

広帯域スペクトルのVALO Grand重合用光照射器は、ISO 10650に準拠した385～515nmの波長範囲で、すべての光硬化製品を重合できるように設計されています。
VALO Grand重合用光照射器は、Ultradent VALO充電式バッテリーおよびバッテリーチャージャーを使用します。
本製品は、標準的な歯科ユニットブラケットに設置できるように設計されていますが、キットに付属のブラケットを使用してカスタムマウントすることもできます。

VALOグランド製品コンポーネント:
•
•
•
•
•
•

1 - VALO Grand重合用光照射器
4 - Ultradent VALO充電式バッテリー
1 - 医療用12VDC AC電源アダプター付きUltradent VALOバッテリー充電器
1 - VALOバリアスリーブサンプルパック
1 - VALO Grandライトシールド
1 - 両面粘着テープ付き重合用光照射器表面取り付けブラケット

コントロールの概要:

VALOコードレス光シールド:

電源ボタン（上または下）凡例

計時ライト

低バッテリー表示

モード/ステータスランプ
タイミング/モード変更ボタン

記載されているあらゆる製品について、ご使用前に指示とSDS情報をすべてよく読み、理解してください。

適応/適用

光活性化歯科用修復材料および接着剤を硬化させるための照明源。

警告および注意事項

1) 重合用光照射器を使用する前にバッテリーを充電してください（「バッテリーのメンテナンス」の項を参照）。
2) 毎回使用する前に、新しいバリアスリーブを重合用光照射器の上に置き、最良の結果を得るためにレンズ上のしわを最小限に抑えてください。。。

注:

VALOコードレス光シールドは楕円形で、回転させると最大限に使用でき、透明なバリアスリーブと共に使用することができます。

ボタンとライト(上)

各段階の手順準備• 二次汚染を防ぎ、歯科用コンポジット材がレンズやワンド本体の表面に付着するのを防ぐため、VALO重合用光照射器の使用ごとに、Ultradentが承認したバリアスリーブを使用する必要があります。
バリアスリーブは、患者1人用です。

写真

(下)

• 機器の損傷を避けるため、金接点やバッテリーコンパートメントのいかなる部分も清掃しないでください。
懸念がある場合は、Ultradentカスタマー・サービスに連絡してください。
• 交差汚染のリスクを防ぐため、
バリアスリーブは1患者1回のみ使用してください。
• 腐食のリスクを減らすため、使用後はバリアスリーブを取り外してください。
• 樹脂の硬化不足のリスクを減らすため、レンズが損傷している場合は重合用光照射器を使用しないでください。

リスクグループ2

本製品から発生する紫外線に気を付けてください。
暴露されると、眼または皮膚が刺激される可能性があります。
適切な保護具を使用してください。
この製品から放射される可能性のある危険な光放射に気を付けてください。
手術灯を見つめないでください。
目に害を及ぼす可能性があります。

• 光源を直視しないでください。
患者、臨床医、助手は、光照射器を使用する際には、必ず琥珀色のUV保護メガネを着用してください。
• 感電の危険を避けるため、この装置の改造は禁止されている。
付属のUltradent VALO電源およびプラグアダプターのみを使用してください。
これらの部品が破損した場合は、使用せず、Ultradentカスタマー・サービスに連絡して交換品を注文してください。
• ポータブルRF通信機器を、30cm（12インチ）
より近くで使用すると性能が低下することがあります。
• 不適切な操作、電磁放射の増加、電磁波耐性の低下を防止するため、認定されたアクセサリー、
ケーブル、電源のみを使用してください（「電磁放射」の項を参照）。
• バッテリーの取り扱いに伴う電気火災の危険を避ける：
⸰ 充電式電池と非充電式電池や他の種類の電池を混ぜないでください。
⸰ 充電式以外のバッテリーを充電しないでください。
充電式でないバッテリーを充電すると、人身事故や物損事故を招く恐れがあります。
⸰ バッテリー、
バッテリー接点、充電器、AC 電源アダプタにオートクレーブをかけたり、液体を吹き付けたりしないでください。
充電器の接点に腐食が見られる場合は、Ultradentカスタマーサービスに連絡し、交換品を注文してください。
⸰ 可燃物の近くでバッテリーを充電しないでください。
⸰ 充電器を手術室に置かないでください。
していたり、へこんでいたり、臭いや液体が出たり、包装が破れていたり、欠けていたり、
その他損傷しているバッテリーは使• 傷害の危険を避けるため、腐食（錆）用しないでください。
交換用電池のご注文は、Ultradent カスタマーサービスまでお電話ください。
• 推奨バッテリーのみを使用してください。
代替バッテリーは故障の原因となります。
どのモードでも口腔内の軟組織を10秒以上近接させないでください。
より長い硬化時間が必• 熱刺激や傷害の危険を避けるため、硬化サイクルの連続は避け、要な場合は、複数の硬化サイクルを使用し、サイクル間に休息時間を設けるか、デュアルキュア製品を使用して軟組織に加熱しないようにしてください。
• 光生物学的副作用や過敏症を患っている患者、化学療法を受けている患者、光感作性薬剤による治療を受けている患者を治療する場合は注意してください。
• 本機は強い磁場や静電気の影響を受けやすく、
プログラミングを妨害する可能性があります。
このような現象が発生したと思われる場合、本機のプラグをしばらく抜いてから、再度コンセントに差し込んでください。
オートクレーブにかけたり、超音波槽、消毒液、洗浄液、液体に浸したりしないでください。
付属の処• 苛性洗剤や研磨剤入りのクリーナーで光照射器を拭いたり、
理指示に従わない場合、重合用光照射器が動作しなくなることがあります。
その他の物を入れないでください。
• 装置の損傷を避けるため、重合用光照射器のバッテリーコンパートメントに指や器具、
270

使用

1) 各パワーモードは、光開始剤を用いた歯科材料の硬化に使用されます。
推奨硬化時間についてはクイックモードガイドをご覧ください。
注：硬化ライトは、スタンダードパワー、
ハイパワープラス、
エクストラパワーモードの順に切り替わるようプログラムされています。
例えば、スタンダードパワー力モードからエクストラパワーモードに変更するには、
ハイパワーモードプラスに切り替えてからエクストラパワーモードに切り替える必要があります。
2) 硬化ライトは、最も直近に使用された計時インターバルとモードを保存します。
モードを変更するか、電池を取り外すと、
デフォルトに戻ります。

操作
硬化モード：スタンダードパワーモード

計時インターバル：5秒、10秒、15秒、20秒。
• 硬化ライトは初めて電源を入れると、このモードになります。
モード/ステータスランプが緑色になり、4つの緑色の計時ランプが点灯して、スタンダードパワーを示します。
• 計時インターバルを変更するには、時間/モードボタンをすばやく押してください。
• 電源ボタンを押して硬化します。
計時インターバルが終了する前に硬化をやめるには、もう一度電源ボタンを押してください。

硬化モード：ハイパワープラスモード

計時インターバル：1秒、2秒、3秒、4秒。
• スタンダードパワーモードから、時間/モード変更ボタンを2秒間押し続けて放します。
モード/ステータスランプがオレンジ色になり、4つの緑色の計時ランプが点灯して点滅し、ハイパワーモードを示します。
• 計時インターバルを変更するには、時間/モードボタンをすばやく押してください。
• いずれかの電源ボタンを押して硬化します。
計時インターバルが終了する前に硬化を止めるには、もう一度電源ボタンを押してください。
• スタンダードパワーモードに戻すには、時間/モード変更ボタンを2秒間押し続けて放すと、エクストラパワーモードに切り替わります。
もう一度2秒間押し続けて放します。
モード/ステータスランプが緑色になり、4つの緑色の計時ランプが点灯して、スタンダードパワーモードになっていることを示します。

硬化モード：エクストラパワーモード

計時インターバル：3秒のみ（注：エクストラパワーモードでは、各硬化サイクルの終わりに2秒の安全遅延が設定されており、連続して硬化を行う際に熱の放出を制限します。
遅延が終わるとビープ音が鳴り、装置は継続的に使用できる状態になります）。
• スタンダードパワーモードから、時間/モード変更ボタンを2秒間押してから、もう一度2秒間押します。
モード/ステータスランプがオレンジ色で点滅し、緑色の計
時ランプのうち3つが点灯して点滅し、エクストラパワーモードを示します。
• 電源ボタンを押して硬化します。
計時インターバルが終了する前に硬化をやめるには、もう一度電源ボタンを押してください。
• スタンダードパワーモードに戻るには、時間/モードボタンを2秒間押してください。
モード/ステータスランプが緑色になり、緑色の計時ランプが点灯して、スタン
ダードパワーモードを示します。
スリープモード：硬化ライトは60秒間使用しないとスリープモードに入り、
モード/ステータスランプがゆっくり点滅します。
硬化ライトは取り上げるか触ると起動し、最後に使用した設定に自動的に戻ります。
電池の寿命を最大限に延ばすため、使用していないときは硬化ライトをいじらないでください。

クリーンアップ

1) 各患者さんで使用した後、使用済みのバリアスリーブは標準廃棄物として処分してください。
2)「処理」セクションを参照してください。

取り付けブラケットの取扱説明

1) ブラケットは平らな、オイルの付いていない表面に取り付ける必要があります。
2) アルコールで表面を拭いてください。
3) ブラケットの粘着テープをはがします。
4) 取り外したときに硬化ライトが上に上がるようにブラケットを配置します。
しっかりと所定の位置に押し込みます。

271

クイックモードガイド:

スタンダードパワー
1000 mW/cm2

モード

電源ボタン

ハイパワープラス
1600 mW/cm2

エクストラパワー
3200 mW/cm2

モード/タイミングLED
時間ボタン
時間オプション

5秒

時間を変更するにはモードを変更するには
凡例

10秒

15秒

20秒

1秒

3秒

3秒のみ

LEDが点滅重合モード

パワー*(mW)

放射照度*(mW/cm2)

総露光時間（秒）ハイパワープラス

1,620

1,500

1/2/3/4

標準

4秒

時間ボタンを2秒間押し続けます。
硬化ライトは次のモードに切り替わります。
LEDが点灯クイック硬化ガイド:

2秒

時間ボタンを短く押して離すと、時間オプションが順番に表示されます。

970

エクストラ

900

2,260

2,100

エネルギー (ジュール）

5/10/15/20

4.8J/9.7J/14.5J/19.4J

3

6.8J

1.6J/3.2J/4.8J/6.5J*レンズ先端からコンポジットの上面まで 2 mm の距離での公称出力。
**サイクルあたりの総エネルギー (ジュール) セクションの値は、小数点以下を四捨五入しています

快速警告指南:電力レベルに関する警告
電池を交換する• 電池残量低下：ゆっくり点滅• シャットオフ：ビープ音3回、点滅• 操作禁止温度に関する警告

クールダウンを行う• ビープ音3回• ゆっくり点滅• 操作禁止サービスに関する警告修理のためにカスタマーサービス
に連絡する
• 無音
• 点滅、2秒• 操作を許可

LEDに関する警告

修理のためにカスタマーサービスに連絡する• 連続したビープ音3回• 1回の急速なフラッシュ• 操作禁止1) レンズ表面に歯科用コンポジットが付着するのを防ぐため、
バリアスリーブを使用してください。
必要であれば、
プラスチック製またはステンレス製の歯科器具を使用して、付着したコンポジットレジストを慎重に除去してください。
レンズを傷つけるような工具は使用しないでください。
2) 光度計は大きく異なり、特定のライトガイドチップとレンズ用に設計されています。
ウルトラデントでは、標準パワーモードでの出力を定期的にチェックすることを推奨しています。
注：一般的な光度計や重合用光照射器のカスタムLEDパックの不正確さにより、真の数値出力は歪みます。
バッテリーのメンテナンス
バッテリーの充電と交換
充電式リン酸鉄リチウム電池4個付き。
ウルトラデントでは、VALO充電式バッテリーのみの使用を推奨しています。

バッテリーの充電方法
1) 充電器とバッテリーは、液体や消毒スプレーがかからない場所に置いてください。
2) プラス (+) 側を充電器のインジケータライトに向けて、バッテリーを充電器に挿入します。
3) 緑色の点灯は、電池が使用可能な状態であることを示します。
4) バッテリーの充電には1～3時間かかります。
バッテリーは、使用準備が整うまで充電器に入れたままにしておいてください。
注：バッテリーを3時間以上充電しても充電器の赤いランプが緑にならない場合は、
バッテリーが劣化（破損）している可能性があり、充電できません。
新しい充電池をお試しいただくか、
ウルトラデント・カスタマー・サービスまでお電話ください。
電池の交換・挿入方法
1) 反時計回りに1/4回転させて、バック キャップを取り外します。
2) バッテリーを取り外します。
3) 新しい電池のプラス（＋）側を先に挿入します。
4) バックキャップの位置を合わせ、時計回りにねじりながら軽く押して、
バックキャップを取り付けます。
キャップは完全に装着されるとカチッと音がします。
5) 本機は使用可能です。

電池残量低下：ローバッテリーインジケータライトが赤色に点滅している場合、電池交換時期です。
電池の残量が少なくなりすぎると、警告音が3回鳴り、電池を再充電するか新しい電池を挿入するまで、重合用光照射器はそれ以上動作しなくなります。
(クイック警告ガイド参照）

充電時間とバッテリー寿命：重合用光照射器のフル充電バッテリー寿命は、モード/時間間隔、
バッテリータイプ、使用量、LED効率によって異なります。
• 推奨充電間隔：必要に応じて、または電池残量インジケータが点灯したら充電してください。
寿命は使用状況により異なります。
• 予備バッテリー：本製品には充電式電池が4本付属しています。
メーカー修理1) 修理は、認可されたサービス担当者のみが行うものとする。
ウルトラデント社は、修理担当者に修理を行うための文書を提供します。
2) 修理、
サービス、校正のためにユニットを送る場合は、必ず重合用光照射器と充電器からバッテリーを取り外してください。
バッテリー、充電器、アダプター、重合用光照射器は、返送用の箱に別々に梱包してください。
3) バッテリーは、地域の規制に従って出荷してください。

保証
Ultradent Products, Inc. (以下「ウルトラデント」) は、本製品を購入日から5年間、本製品に付属の取扱説明書に従って使用した場合、(i) 本製品に付属するウルト
ラデントの文書に記載された仕様にすべての重要な点において適合すること、および (ii) 材料および製造上の欠陥がないことを保証します。

この限定保証は、譲渡不可であり、最初の購入者にのみ適用され、その後の製品の所有者には適用されません。この限定保証は、バッテリー、充電器、アダプター、
アダプティブレンズなどのその他の付属部品には適用されません。
本限定保証は、過失、乱用、誤用、事故、改造、改ざん、または該当する使用説明書に従わなかったことが原因で製品が故障または破損した場合は無効となります。例として、製品を落として破損した場合は本保証の対象外となります。本限定保証の適用を受けるには、購入証明書（販売レシートまたは同様の書類など）を欠陥製品とともにウルトラデントに提出する必要があります。
本書に記載された保証条件に適合する欠陥製品については、
ウルトラデントの独自の裁量により、修理または交換のいずれかを行います。いかなる場合においても、
ウルトラデントの責任は、購入者が支払った購入価格を超えないものとします。いかなる場合においても、
ウルトラデントは、本製品の使用に起因または関連して生じた間接的、偶発的、予見的、不測的、特別または結果的損害について責任を負いません。

処理

本製品はバリアスリーブとの併用を目的としています。
メンテナンスと一般的なクリーニングについては、上記をご参照ください。

保管と廃棄

重合用光照射器を2週間以上保管する場合、または移動用に梱包する場合は、必ずバッテリーを取り外してください。
バッテリーを充電せずに長期間放置すると、機能しなくなったり、充電できなくなったりすることがあります。
バッテリーを40℃を超える場所や直射日光の当たる場所に保管しないでください。

メンテナンス

重合用光照射器の一般的なクリーニング
使用後は毎回、ガーゼか柔らかい布を認可された表面消毒剤で湿らせ、表面とレンズを拭きます。
許可されていないクリーナーは、重合用光照射器に損傷を与える可能性があります。
許容可能なクリーナー：• 70%イソプロピルアルコール• 70%エタノール

ライトシールドの一般的なクリーニング：
VALO Cordlessライトシールドは、任意の表面消毒剤を使用して清掃してください。
オートクレーブは使用しないでください。
修理
ユーザーによるメンテナンス
272

硬化ライトの保管と輸送:• 温度：+10°C～+40°C• 相対湿度：10％～95％• 周囲圧力：500～1060 hPa

電子廃棄物（硬化ライト、充電器、電池、電源など）を処分する際は、各地の廃棄物およびリサイクルのガイドラインに従ってください。

技術的な考慮事項
付属品

アイテム

CE情報

VALOコードレス光シールド
273

アイテム
VALO充電器

CE情報

技術情報/データ

効果的な複合硬化波長帯域:

有効な複合材硬化波長帯(使用可能波長範囲：385～515nm）

VALO電池

有効下限
波長帯395-415 nm

有効上波長帯440-480 nm

エクストラ

ハイパワー

空間放射電力 (mW/nm)

標準

WAVELENGTH (nm)

属性
镜头

波長範囲

情報/仕様

直径11.7 mm• 利用可能な波長範囲：385～515 nm• ピーク波長：395～415 nmおよび440～480 nm

Nominal Radiant Exitance Comparison Chart

測定計器
光強度表

メーターの口径

スタンダードパワー（±10%）

ハイパワープラス（±10％）
エクストラパワー（±10％）‡ Gigahertzスペクトルアナライザー
発散度

合計パワー

900 mW/cm²

970 mW

1500 mW/cm²

1615 mW

15 mm

2100 mW/cm²

VALOグランド硬 定格：IEC 60601-1（安全性）、IEC 60601-1-2
化ライト
（EMC）
充電器用電源274

出力 - 500mAで12VDC
入力 - 100VAC～240VAC
Ultradent P/N 5930 VALO充電器用電源、国際仕様差し込みプラグ付き

15 mm

放射静電容量は、測定器の性能、測定方法、光の配置によって異なります。
Gigahertz-Optik スペクトラムアナライ
ザで測定した場合、放射静電容量は ISO
10650 に準拠します。

2260 mW

重量:• 電池を含む場合:6オンス（190グラム）• 電池を含まない場合:5オンス（150グラム）
寸法：（203 x 32.5 x 27）mm

定格：IEC 60601-1（安全性）
コードの長さ - 1.8メートル
VALO充電器の電源は医療グレードのクラスII電源であり、主電源からの絶縁を提供します。

275

属性

その他の情報情報/仕様

VALO充電器

電磁波に関するガイダンスと製造宣言

VALO 3.6VDCリチウム鉄リン酸スマートバッテリー充電器：• 完全に充電されると自動的にシャットオフ
• 欠陥のある電池を自動検出• 保護:熱、過充電、短絡、逆極性
⸰ 赤色LED – 充電中⸰ 緑のLED - 空または完全に充電⸰ LEDオフ - 短絡
• 充電時間：1〜3時間
定格：CE、WEEE

硬化ライトは下記の電磁環境で操作するよう意図されています。
お客様またはユーザーは、
このような環境で使用されていることを確認する必要があります。

操作条件
デューティサイクル

トラブルシューティング

温度：+10°C～+32°C
相対湿度：10％～95％
周囲圧力：700～1060 hPa

硬化ライトは短期間の操作向けに設計されています。
最高周囲温度（32°C）で1分間バックツーバックサイクリングし、30
分間OFFになります（冷却時間）。

提案されている下記の解決策で問題が修正されない場合は、Ultradent（800.552.5512）までお電話ください。
米国以外では、Ultradent販売店または歯科代理店にご相談ください。
問題

ライトが点灯しない

ライトが希望する時間よりも早く消える

ライトがレジンを適切に硬化しない電池を充電できない

充電器が電池を充電しないモードや計時インターバルを変
更できない

276

準拠

電磁環境 - ガイダンス

RF放射
CISPR 11

クラスB

硬化ライトは、内部機能のためにのみ電気および電磁エネルギーを使用します。
このため、RF放射は非常に低く、近くにある電子機器の干渉の原因になる可能性はほとんどありません。

電圧変動/フリッカー放射IEC 61000-3-3

該当なし

RF放射
CISPR 11

充電式：安全化学リン酸鉄リチウム（LiFePO4）RCR123A• 動作電圧：3.2VDC、400mAH（必要なワット時）
定格UL62133-2、CE、RoHS、WEEE
セルの容量が70%に達するまでの充放電サイクルは2000回以上。
電池は3～5年ごとに交換することをお勧めしますが、使用状況により寿命が異なる場合があります。
必要に応じて交換してください。

VALO電池排出ガス試験

可能な解決策1) 時間/モードまたは電源ボタンを押して、節電モードから復帰します。
2) 電池の充電状態については、赤色の残量低下インジケーターを確認してください。
3) 新しい電池が正しく装置に挿入されていることを確認します。
4) 赤色と黄色の警告LEDが点滅している場合は、硬化ライトの内部温度が安全限度に達しています。
硬化ラ
イトを10分間冷ますか、冷たい濡れたタオルで素早く装置を冷却してください。
5) 赤色の警告LEDが点滅し、ビープ音が連続して鳴った場合は、Ultradentカスタマーサービスに連絡して修理を依頼してください。
1) モードと計時ライトをチェックし、入力された時間が正しいか確認してください。
2) 電池の充電状態については、残量低下インジケーターを確認します。
3) 新しい電池が正しく装置に挿入されていることを確認します。

1) レンズに硬化レジン/合成物が付着していないか確認してください。
2) 適切な琥珀色のUV保護ゴーグルを使用し、LEDライトが機能していることを確認します。
3) ライトメーターのパワーレベルを確認してください。
ライトメーターを使用する場合、Ultradentはスタンダ
ードパワーモードで硬化ライトをチェックするよう推奨します。
注：一般的なライトメーターの不正確さと硬化ライトが使用するカスタムLEDパックのために実際の数値出力
は歪められます。
ライトメーターは大きく異なり、特定のライトガイドチップとレンズ用に設計されています。
4) 硬化レジンの有効期限を確認してください。
5) メーカーが推奨する適切な取り扱い方法に従ってください。

高調波放射IEC 61000-3-21) 硬化ライトのロックが解除されたことを一連のビープ音で示されるまで、時間/モードボタンと電源ボタンの両方を押し続けます。

該当なし

硬化ライトは電池を使用しており、EMI、RF、サージ抑制の影響を受けません。

硬化ライトは電池で駆動するため、高調波放出および電圧変動試験は適用されません。

硬化ライトはあらゆる施設での使用に適しています。
この中には、家庭用施設や、家庭用の建物に供給される公共の低電圧電源ネットワークに直接接続された施設が含まれます。

電磁波放出に関するガイダンスと製造宣言

硬化ライトは下記の電磁環境で操作するよう意図されています。
お客様またはユーザーは、
このような環境で使用されていることを確認する必要があります。
イミュニティ試験

IEC 60601試験レベル

準拠レベル

電気的高速トランジェント/
バースト
IEC 61000-4-4±2 kV（電源ライン）±1 kV（入出力ライン）

電源入力ラインの電圧、低下、ショート、中断、変化<5％U（0.5サイクルのUで95％以上低下）±2 kV（電源ライン）
硬化ライトは電池駆動式なので、AC主電源に
注1：硬化ライトにはI/Oポート 接続することはできません。
がありません硬化ライトは電池駆動式なので、電気的過渡、±1kV（ライン間）
サージ、電圧低下、短絡、中断、AC主電源の変±2kV（ラインからアース））
動の影響を受けません。

静電気放電（ESD）
IEC 61000-4-2

サージ
IEC 61000-4-5

IEC 61000-4-111) 電池が充電器に正しい方向で挿入されていることを確認し、電池を1〜3時間充電してください。
2) 充電器の赤色のランプが緑色に変わらない場合は、Ultradentカスタマーサービスに連絡して、交換用の電池や充電器を注文してください。
3) 充電器の緑色と赤色の両方のランプが見えない場合は、Ultradentカスタマーサービスに連絡して、充電器やACアダプターを注文するか交換してください。
1) 充電器が接続され、正常に機能しているコンセントにACアダプターが差し込まれていることを確認します。
2) 充電器の緑色または赤色のランプが見えない場合は、Ultradentカスタマーサービスに連絡し、新しい充電器やACアダプターを入手してください。

グループ1±8 kV（接点）±15 kV（空気）±1kV（ライン間）±2kV（ラインからアース）

IEC 61000-4-8<5％U（0.5サイクルのUで95％以上低下）

電磁環境ガイダンス

物理的環境は以下に制限してください:
1) IPコード：IP202) 液体に浸さないでください。
3) 可燃性ガスの周囲では使用しないでくださ
い。
装置は非APG、非APです。
4) 保管湿度範囲：10％～95％
5) 保管温度範囲：10°C～40°C

備品の電池充電器は上記の影響を受ける場合がありますが、VALOコードレス硬化ライトの操作とは別であり、重要ではありません。

40％U
40％U
硬化ライトの電池の電圧が4VDCに低下する（5サイクルのUで60％低下） （5サイクルのUで60％低下）と、装置は操作できません。
VALOコードレス硬化ライトはオフになります。
新しい電池を入れて70％U
70％U
適切な電力レベルに戻すと、硬化ライトは再起（25サイクルのUで30％（25サイクルのUで30％動し、電源喪失前と同じ状態に戻ります。
硬化
低下）
低下）
ライトは電源喪失の場合、自己復旧します。
<5％U（5秒間のUで95％以上
低下）

電源周波数
（50/60Hz）磁界±8 kV（接点）±15 kV（空気）30 A/m<5％U（5秒間のUで95％以上
低下）
注2：自己回復
30 A/m

電力周波数磁場は、通常の住宅、在宅医療、商業、病院または軍事環境における典型的な場
所の特徴的なレベルでなければなりません。

注1：硬化ライトにはポートやアクセス可能なI/Oラインは装備されていません。

277

無生物支援システム向けの電磁波放出に関するガイダンスと製造宣言

硬化ライトは下記の電磁環境で操作するよう意図されています。
お客様またはユーザーは、
このような環境で使用されていることを確認する必要があります。
イミュニティ試験

IEC 60601試験レベル 準拠レベル

伝導RF

3 Vrms3 VrmsIEC 61000-4-6

150 kHz～80 MHz

150 kHz～80 MHz

放射RF

3 V/m

3 V/m

電磁環境ガイダンス

ポータブルおよび携帯RF通信機器は、ケーブルを含む硬化ライトのどの部分に対しても、送信機の周波数に適用される式から計算された推奨分離距離より近くで使用しないでください。
推奨分離距離:80 MHz～800 MHzแสงบ่่ม

รายละเอีียดผลิิตภััณฑ์์

ด้้วยสเปกตรััมบรอดแบนด์์ เครื่่�องบ่่มแสง VALO Grand ได้้รับั การออกแบบมาเพื่่�อรวมตััวกัับผลิิตภััณฑ์์บ่ม่ ด้้วยแสงทั้้�งหมดในช่่วงความยาวคลื่่�น 385-515 นาโนเมตรต่่อ ISO 10650 เครื่่�องบ่่มแสง VALO Grand ใช้้แบตเตอรี่่� Ultradent VALO แบบชาร์์จได้้ และเครื่่�องชาร์์จแบตเตอรี่่� ไฟบ่่มได้้รับั การออกแบบให้้วางอยู่่ใ� นขายึึดชุุดทัันตกรรมมาตรฐาน หรืือสามารถติิดตั้้�งแบบกำำ�หนดเองได้้โดยใช้้ขายึึดที่่�มาพร้้อมกัับชุุดอุุปกรณ์์

ส่่วนประกอบผลิิตภััณฑ์์ VALO แกรนด์์:
•
•
•
•
•
•

1 – เครื่่�องบ่่มแสง VALO Grand4 – แบตเตอรี่่� Ultradent VALO แบบชาร์์จได้้
1 – ที่่�ชาร์์จแบตเตอรี่่� Ultradent VALO พร้้อมอะแดปเตอร์์ไฟ AC เกรดการแพทย์์ 12VDC
1 – ชุุดตััวอย่่างปลอกกั้้�น VALO
1 – ที่่�บังั แสง VALO Grand1 – ตััวยึึดยึดพื้้ึ � นผิิวเครื่่�องบ่่มแสงด้้วยเทปกาวสองหน้้า

ภาพรวมของส่่วนควบคุุม:ภาพถ่่าย

800 MHz～2.5 GHzปุ่่�มและแสงไฟ

80 MHz～2.5 GHz

80 MHz～2.5 GHz

硬化ライトは、放射されたRF妨害が制御される電磁環境で操作するよう意図されています。硬化ライトのユーザーは、通信機器の最大出力電力に応じて、以下に推奨されているポータブル・携帯RF通信機器（送信機）と硬化ライト間の最短距離を維持することによって電磁干渉を防止できます。
150 kHz～80 MHz

0.01
1

0.1
10

100

送信機の周波数による分離距離（メートル）
80MHz～800MHz

800MHz～2.5GHz

0.12メートル

0.035メートル

0.07メートル

1.7メートル

0.35メートル

11.7メートル

3.5メートル

0.37メートル
3.7メートル

0.11メートル

1.11メートル

0.22メートル
0.7メートル

2.22メートル
7.0メートル

3Vrmsの値はV1に硬化ライトは、IEC 60601-1-2：2014に従って試験され、80 MHz〜2.5 GHzの間で10 V/mの放射磁界強度で合格しています。
対応し、10V/mの値は上記の式のE1に対応します。
上記に含まれていない最大出力定格の送信機の場合、推奨分離距離（d）（メートル、m）は送信機の周波数に適用される式を使用して推定できます。
ここで、「P」は送信機製造業者の指定した送信機の最大出力定格（ワット、W）です。
注1：80 MHzと800 MHzでは、より高い周波数範囲の分離距離が適用されます。
注2：これらのガイドラインは、
すべてのケースに適用できないことがあります。電磁波伝播は、構造物、物体、人からの吸収と反射の影響を受けます。
278ตััวบ่่งชี้้� - แบตเตอรี่่อ่� ่อนไฟแสดงโหมด/สถานะ

次の記号が付いている機器の近くでは干渉が発生することがあります:

ポータブルおよび携帯RF通信機器とVALOコードレス硬化ライトの間の推奨分離距離に関するガイダンスおよび製造業者の宣言

定格最大出力パワー
定格最大出力（Pはワット）ไฟเวลา

電磁サイト調査aによって決定される固定RF送信機からの電磁場の強さは、各周波数範囲bの準拠レベル未満でなくてはなりません。

注1：80MHzおよび800MHzでは、より高い周波数範囲が適用されます。
注2：これらのガイドラインは、
すべてのケースに適用できないことがあります。電磁波伝播は、構造物、物体、人からの吸収と反射の影響を受けます。
a
固定送信機（無線（携帯・コードレス）電話、陸上移動無線、
アマチュア無線、AMおよびFMラジオ放送、テレビ放送の基地局など）からの場の強さは、理論的に正確に予測できません。
固定RF送信機に起因する電磁環境を評価するには、電磁場調査を考慮する必要があります。
硬化ライトが使用される場所で測定された磁界の強さが該当する上記のRF準拠レベルを超える場合は、VALOコードレス硬化ライトを観察して正常な操作を検証する必要があります。異常な動作が観察された場合は、硬化ライトの向きを変えるか再配置するといった追加措置が必要な場合があります。
b
150 kHz〜80 MHzの周波数範囲を超える場合、場の強さは3 V/m未満でなくてはなりません。

คำำ�อธิิบาย

ปุ่่�มเปิิด/ปิิด(ด้้านบนหรืือด้้านล่่าง)「P」は送信機製造業者の指定した最大出力定格（ワット、W）で、「d」は推奨される分離距離（m）です。
IEC 61000-4-3

TH

VALO™ แกรนด์์

ปุ่่�มเปลี่่ย� นเวลา/โหมด(ด้้านล่่าง)(ด้้านบน)สำำ�หรัับผลิิตภััณฑ์์ทั้้�งหมดที่่�อธิิบายไว้้ ให้้อ่่านและทำำ�ความเข้้าใจคำำ�แนะนำำ�และข้้อมููล SDS ทั้้�งหมดก่่อนใช้้งาน

ตััวบ่่งชี้้สำ� �ำ หรัับการใช้้ / วััตถุุประสงค์์ที่่�กำำ�หนดไว้้

แหล่่งที่่�มาของแสงสว่่างสำำ�หรัับวััสดุุบููรณะและสารยึึดติิดที่่�ใช้้แสงบ่่ม

คำำ�เตืือนและข้้อควรระวััง
กลุ่่�มความเสี่่�ยง 2ข้้อควรระวััง ผลิิตภััณฑ์์นี้้ป� ล่่อยแสง UV การเผยผึ่่�งอาจก่่อให้้เกิิดการระคายเคืืองตาหรืือผิิวหนััง ใช้้การป้้องกัันที่่�เหมาะสม
ข้้อควรระวััง ผลิิตภััณฑ์์นี้้อ� าจมีีการแผ่่รังั สีีแสงที่่�เป็็นอัันตราย อย่่าจ้้องที่่�หลอดไฟ อาจเป็็นอัันตรายต่่อดวงตา
• อย่่ามองไปที่่�เอาต์์พุุตแสงโดยตรง ผู้้�ป่่วย แพทย์์ และผู้้�ช่่วยควรสวมอุุ ปกรณ์์ป้้องกัันดวงตา UV สีีอำำ�พันั เสมอเมื่่�อใช้้เครื่่�องบ่่มแสง
• เพื่่�อป้้องกัันความเสี่่ย� งจากไฟฟ้้าช็็อต ไม่่อนุุญาตให้้ทำำ�การดััดแปลงอุุ ปกรณ์์นี้้� ใช้้เฉพาะแหล่่งจ่่ายไฟ Ultradent VALO และอะแดปเตอร์์ปลั๊๊�กที่่�ให้้มาเท่่านั้้�น หากส่่วนประกอบเหล่่านี้้�เสีียหาย อย่่าใช้้และติิดต่่อฝ่่ายบริิการ
ลููกค้้าของ Ultradent เพื่่�อสั่่�งซื้้�อชิ้้�นส่่วนทดแทน
• อุุ ปกรณ์์สื่�อ่ สาร RF แบบพกพาอาจลดประสิิทธิิภาพหากใช้้ใกล้้เกิิน 30 ซม. (12 นิ้้�ว)• ใช้้เฉพาะอุุ ปกรณ์์เสริิม สายไฟ และแหล่่งจ่่ายไฟที่่�ได้้รับั อนุุญาตเท่่านั้้�น เพื่่�อป้้องกัันการทำำ�งานที่่�ไม่่เหมาะสม เพิ่่�มการปล่่อยคลื่่�นแม่่เหล็็กไฟฟ้้า หรืือภููมิิคุ้้�มกัันแม่่เหล็็กไฟฟ้้าลดลง (โปรดดููส่่วนการปล่่อยคลื่่�นแม่่เหล็็กไฟฟ้้า)• เพื่่�อหลีีกเลี่่ย� งความเสี่่ย� งจากไฟไหม้้จากไฟฟ้้าที่่�เกี่่�ยวข้้องกัับการจััดการแบตเตอรี่่:�• อย่่าใช้้ร่ว่ มแบตเตอรี่่แ� บบชาร์์จซ้ำำ��ได้้กัับแบตเตอรี่่ที่� ่�ไม่่สามารถชาร์์จซ้ำำ��ได้้หรืือแบตเตอรี่่ป� ระเภทอื่่�น
• อย่่าพยายามชาร์์จแบตเตอรี่่ที่� ่�ไม่่สามารถชาร์์จใหม่่ได้้ การชาร์์จแบตเตอรี่่ที่� ่�ไม่่สามารถชาร์์จซ้ำำ��ได้้อาจส่่งผลให้้เกิิดการบาดเจ็็บหรืือทรััพย์สิ์ นิ เสีียหาย
• ห้้ามนึ่่�งหรืือฉีีดพ่่นแบตเตอรี่่� หน้้าสััมผััสของแบตเตอรี่่� ที่่�ชาร์์จ หรืืออะแดปเตอร์์ไฟ AC ด้้วยของเหลวทุุกชนิิด หากเกิิดการกััดกร่่อนบนหน้้าสััมผััสของที่่�ชาร์์จแบตเตอรี่่� โปรดติิดต่่อฝ่่ายบริิการลููกค้้าของ Ultradent เพื่่�อ
สั่่�งอุุ ปกรณ์์เปลี่่ย� นใหม่่
• อย่่าชาร์์จแบตเตอรี่่ใ� กล้้กัับวััสดุุไวไฟ
• อย่่าเก็็บอุุ ปกรณ์์ชาร์์จไว้้ในห้้องการปฏิิบัติั ิงานทางคลิินิิก
• เพื่่�อหลีีกเลี่่ย� งความเสี่่ย� งต่่อการบาดเจ็็บ ห้้ามใช้้แบตเตอรี่่ที่� ่�สึกึ กร่่อน (เป็็นสนิิม) มีีรอยบุุบ มีีกลิ่่�นหรืือของเหลว มีีห่่อฉีีกขาดหรืือขาดหายไป หรืือเสีียหายอย่่างอื่่�น ติิดต่่อฝ่่ายบริิการลููกค้้าของ Ultradent เพื่่�อสั่่�ง
แบตเตอรี่่ทด� แทน• ใช้้แบตเตอรี่่ที่� ่�แนะนำำ�เท่่านั้้�น แบตเตอรี่่ท� างเลืือกอื่่�นๆอาจทำำ�ให้้เกิิดการทำำ�งานผิิดพลาดได้้
• เพื่่�อป้้องกัันความเสี่่ย� งของการระคายเคืืองหรืือการบาดเจ็็บจากความร้้อน ให้้หลีีกเลี่่ย� งการบ่่มแสงแบบติิดกััน และอย่่าให้้ทิิชชู่่ใ� นช่่องปากอยู่่ใ� กล้้กัันนานกว่่า 10 วิินาทีีในโหมดใดก็็ตาม หากต้้องใช้้เวลาในการบ่่มนานขึ้้�น ให้้ใช้้รอบการบ่่มหลายรอบโดยมีีช่่วงเวลาพัักระหว่่างรอบ หรืือใช้้ผลิิตภััณฑ์์แบบ dual-cure เพื่่�อหลีีกเลี่่ย� งไม่่ให้้เนื้้�อเยื่่�ออ่่อนร้้อน
• ใช้้ความระมััดระวัังในการรัักษาผู้้�ป่่วยที่่�มีีอาการไม่่พึงึ ประสงค์์จากปฏิิกิิริยิ าหรืือความไวต่่อแสงทางชีีวภาพ ผู้้�ป่่วยที่่�กำำ�ลังั รัับการรัักษาด้้วยเคมีีบำำ�บัดั หรืือผู้้�ป่่วยที่่�ได้้รับั การรัักษาด้้วยยาไวแสง
• อุุ ปกรณ์์นี้้อ� าจไวต่่อสนามแม่่เหล็็กหรืือไฟฟ้้าสถิิตที่่�รุุนแรง ซึ่่�งอาจรบกวนการตั้้�งโปรแกรมได้้ หากคุุณสงสััยว่่าสิ่่�งนี้้�เกิิดขึ้้�น ให้้ถอดปลั๊๊�กเครื่่�องชั่่�วคราวแล้้วเสีียบกลัับเข้้าไปในเต้้ารัับอีีกครั้้�ง
• ห้้ามเช็็ดเครื่่�องบ่่มแสงด้้วยน้ำำ��ยาทำำ�ความสะอาดที่่�มีีฤทธิ์์�กััดกร่่อนหรืือสารที่่�กััดกร่่อน เครื่่�องนึ่่�งฆ่่าเชื้้�อ หรืือจุ่่�มลงในอ่่างอััลตราโซนิิก สารฆ่่าเชื้้�อ น้ำำ��ยาทำำ�ความสะอาด หรืือของเหลวทุุกชนิิด การไม่่ปฏิิบัติั ิตามคำำ�แนะนำำ�ใน
การประมวลผลที่่�ให้้มาอาจทำำ�ให้้แสงบ่่มไม่่สามารถทำำ�งานได้้
• เพื่่�อหลีีกเลี่่ย� งไม่่ให้้อุุปกรณ์์เสีียหาย ห้้ามสอดนิ้้�ว เครื่่�องมืือ หรืือวััตถุุอื่� ่น ๆ เข้้าไปในช่่องใส่่แบตเตอรี่่ข� องไฟบ่่ม
• เพื่่�อหลีีกเลี่่ย� งไม่่ให้้อุุปกรณ์์เสีียหาย อย่่าพยายามทำำ�ความสะอาดหน้้าสััมผััสสีีทองหรืือส่่วนใด ๆ ของช่่องใส่่แบตเตอรี่่� ติิดต่่อฝ่่ายบริิการลููกค้้าของ Ultradent หากมีีข้้อกัังวล
• เพื่่�อป้้องกัันความเสี่่ย� งของการปนเปื้้� อนข้้าม ปลอกกั้้�นจึึงมีีไว้้ใช้้สำ�ำ หรัับผู้้�ป่่วยรายเดีียว
279

• เพื่่�อลดความเสี่่ย� งต่่อการกััดกร่่อน ให้้ถอดปลอกกั้้�นออกหลัังการใช้้งาน
• เพื่่�อลดความเสี่่ย� งของเรซิินที่่�บ่ม่ น้้อยเกิินไป อย่่าใช้้แสงบ่่มหากเลนส์์เสีียหาย่่

ข้้อแนะนำำ�เกี่่�ยวกัับโหมดแบบย่่อ:High Power Plus
1600 mW/cm2

พลัังงานมาตรฐาน
1000 mW/cm2

โหมดคำำ�แนะนำำ�เกี่่�ยวกัับขั้้�นตอน
การจััดเตรีียม

Xtra Power
3200 mW/cm2

ปุ่่�มเปิิด/ปิิด1) ชาร์์จแบตเตอรี่่ก่� ่อนการใช้้ไฟบ่่ม (ดููหััวข้้อการดููแลรัักษาแบตเตอรี่่)�
2) ก่่อนการใช้้งานแต่่ละครั้้�ง ให้้วางปลอกกั้้�นใหม่่ไว้้เหนืือแสงบ่่ม เพื่่�อลดรอยยัับบนเลนส์์เพื่่�อให้้ได้้ผลลััพธ์ที่์ ่�ดีีที่่�สุดุ

ไฟ LED โหมด/ระยะเวลา

บัันทึึก:• การใช้้ปลอกกั้้�นที่่�ถููกสุุขอนามััยจะช่่วยลดปริิมาณแสงได้้ 5-10% เนื่่�องจากกำำ�ลังั ขัับสููงของแสงบ่่ม จึึงทำำ�ให้้ได้้การบ่่มที่่�เพีียงพอ
• ต้้องทำำ�ความสะอาดและฆ่่าเชื้้�อแสงบ่่มด้้วยสารทำำ�ความสะอาดและ / หรืือสารฆ่่าเชื้้�อที่่�เหมาะสมหลัังจากใช้้กัับผู้้�ป่่วยแต่่ละราย ดููหััวข้้อการใช้้งาน

ปุ่่�มเวลา

แผ่่นกัันแสงไร้้สาย VALO:

แผ่่นกั้้�นแสงไร้้สาย VALO เป็็นรููปวงรีี สามารถหมุุนได้้เพื่่�อการใช้้งานสููงสุุดและสามารถใช้้กัับปลอกหุ้้�มแบบโปร่่งใสได้้
5วิินาทีี

ตััวเลืือกเวลา

การใช้้

10วิินาทีี

15วิินาทีี

20วิินาทีี

1วิินาทีี

1) โหมดพลัังงานแต่่ละโหมดใช้้สำ�ำ หรัับการบ่่มวััสดุุทางทัันตกรรมที่่�มีีแสงเป็็นตััวบ่่งชี้้� ดููคำำ�แนะนำำ�โหมดแบบสั้้�นเพื่่�อดููเวลาการบ่่มที่่�แนะนำำ�
หมาย: แสงบ่่มจะถููกตั้้�งโปรแกรมให้้หมุุนเวีียนจากโหมดพลัังงานมาตรฐาน ไปเป็็น พลัังงานสููงเพิ่่�ม และไปยัังโหมดพลัังงานสููงพิิเศษ ตามลำำ�ดัับ ตััวอย่่างเช่่น หากต้้องการเปลี่่ย� นจากโหมดพลัังงานมาตรฐานเป็็นโหมดพลัังงาน
สููงพิิเศษ จำำ�เป็็นต้้องหมุุนเวีียนเข้้าสู่่โ� หมดพลัังงานสููงเพิ่่�ม และจากนั้้�นจึึงเปลี่่ย� นเป็็นโหมดพลัังงานสููงพิิเศษ
2) แสงบ่่มจะจััดเก็็บช่่วงเวลาและโหมดที่่�ใช้้ล่า่ สุุด และจะเริ่่�มกลัับมาที่่�โหมดนี้้�เมื่่�อใดก็็ตามที่่�มีีการเปลี่่ย� นแปลงโหมดหรืือหากถอดแบตเตอรี่่อ� อกในการเปลี่่ย� นเวลา

กดแล้้วปล่่อยปุ่่�มเวลาอย่่างรวดเร็็วเพื่่�อวนผ่่านตััวเลืือกเวลา

ในการเปลี่่ย� นโหมด

กดปุ่่�ม Time ค้้างไว้้ 2 วิินาทีีแล้้วปล่่อย แสงบ่่มจะวนไปสู่่โ� หมดถััดไป

การทำำ�งาน
โหมดการบ่่ม: โหมดพลัังงานมาตรฐาน

คำำ�อธิิบาย

ไฟ LED นิ่่�ง

ช่่วงระยะเวลา: 5, 10, 15, 20 วิินาทีี
• แสงบ่่มจะเริ่่�มต้้นที่่�โหมดนี้้�เมื่่�อเปิิดใช้้งานครั้้�งแรก ไฟสถานะ / โหมดจะเป็็นสีีเขีียวและไฟสััญญาณตั้้�งค่่าเวลาสีีเขีียวสี่่ด� วงสว่่างขึ้้�น ซึ่่�งระบุุว่า่ เป็็นโหมดพลัังงานมาตรฐาน
• ในการเปลี่่ย� นช่่วงระยะเวลา ให้้กดปุ่่�ม Time/Mode อย่่างรวดเร็็ว
• กดปุ่่�ม Power เพื่่�อบ่่ม หากต้้องการหยุุดการบ่่มก่่อนที่่�ช่ว่ งระยะเวลาที่่�กำำ�หนดจะเสร็็จสิ้้�นลง ให้้กดปุ่่�ม Power อีีกครั้้�ง

4วิินาทีี

3 วิินาทีีเท่่านั้้�น

ไฟ LED กระพริิบ

โหมดการฉาย

กำ�ลงั ไฟฟ้ า * (มิลลวิ ตั ต์)การฉายรังสี * (mW/cm2)

เวลาเปิ ดรบั แสงทั้งหมด (วินาที)พลังงาน** (จูลส์)มาตรฐาน

970

900

5/10/15/20

4.8J/9.7J/14.5J/19.4J

High Power Plus

1,620

1,500

1/2/3/4

1.6J/3.2J/4.8J/6.5J

Xtra

2,260

2,100

3

6.8J

โหมดการบ่่ม: โหมดพลัังงานสููงเพิ่่�ม

ช่่วงระยะเวลา: เพีียง 3 วิินาทีีเท่่านั้้�น (หมายเหตุุ: โหมดพลัังงานสููงพิิเศษมีีการหน่่วงเวลาเพื่่�อความปลอดภััย 2 วิินาทีีในตอนท้้ายของแต่่ละรอบการบ่่ม เพื่่�อจำำ�กััดความร้้อนในระหว่่างการบ่่มต่่อเนื่่�อง ในช่่วงท้้ายของการหน่่วง
เวลา เสีียงบี๊๊�บจะแจ้้งว่่าเครื่่�องพร้้อมสำำ�หรัับการทำำ�งานต่่อเนื่่�อง)• จากโหมดพลัังงานมาตรฐาน กดปุ่่�มเปลี่่ย� นเวลา / โหมด เป็็นเวลา 2 วิินาทีีแล้้วปล่่อย และกดอีีกครั้้�งเป็็นเวลา 2 วิินาทีีแล้้วปล่่อย ไฟสถานะ / โหมดจะเป็็นสีีส้้มและกระพริิบและไฟจัับเวลาสีีเขีียวสามดวงจะติิดสว่่างและกระ
พริิบซึ่่�งระบุุโหมดพลัังงานสููงพิิเศษ
• กดปุ่่�ม Power เพื่่�อบ่่ม หากต้้องการหยุุดการบ่่มก่่อนที่่�ช่ว่ งเวลาที่่�กำำ�หนดจะเสร็็จสิ้้�น ให้้กดปุ่่�ม Power อีีกครั้้�ง
• หากต้้องการกลัับไปที่่�โหมดพลัังงานมาตรฐานให้้กดปุ่่�ม เวลา / โหมด ค้้างไว้้ 2 วิินาทีีแล้้วปล่่อย ไฟสถานะ / โหมด จะเป็็นสีีเขีียวและไฟสััญญาณบอกเวลาสีีเขีียวจะติิดสว่่างเพื่่�อระบุุโหมดพลัังงานมาตรฐาน

3วิินาทีี

คู่่�มืือการบ่่มแบบย่่อ:ช่่วงระยะเวลา: 1, 2, 3, 4 วิินาทีี
• จากโหมดพลัังงานมาตรฐาน กดปุ่่�มเปลี่่ย� นเวลา / โหมด ค้้างไว้้ 2 วิินาทีีแล้้วปล่่อย ไฟสถานะ / โหมดจะเป็็นสีีส้้มและไฟกำำ�หนดเวลาสีีเขีียวสี่่ด� วงจะติิดสว่่างและกระพริิบซึ่่�งระบุุโหมดพลัังงานสููง
• หากต้้องการเปลี่่ย� นช่่วงเวลา ให้้กดปุ่่�ม เวลา / โหมด อย่่างรวดเร็็ว
• กดปุ่่�มเปิิด/ปิิด เพื่่�อบ่่มตััว ในการหยุุดการบ่่มก่่อนที่่�จะเสร็็จสิ้้�นช่่วงเวลา ให้้กดปุ่่�มเปิิด/ปิิดอีีกครั้้�ง
• หากต้้องการกลัับสู่่โ� หมดพลัังงานมาตรฐานให้้กดปุ่่�ม เปลี่่ย� นเวลา / โหมด ค้้างไว้้เป็็นเวลา 2 วิินาทีีแล้้วปล่่อย ซึ่่�งจะวนไปยัังโหมดพลัังงานพิิเศษ กดค้้างไว้้อีีก 2 วิินาทีีแล้้วปล่่อย ไฟสถานะโหมด / จะเป็็นสีีเขีียวและไฟ
สััญญาณบอกเวลาสีีเขีียวสี่่ด� วงสว่่างขึ้้�นซึ่่�งระบุุว่า่ เป็็นโหมดมาตรฐาน

โหมดการบ่่ม: โหมดพลัังงานสููงพิิเศษ

2วิินาทีี

*เอาต์พุตปกติที่ระยะห่าง 2 มม. จากปลายเลนส์ถึงพื้นผิวด้านบนของคอมโพสิต
**ค่าในส่วนพลังงานทั้งหมดต่อรอบ (จูล) จะถูกปั ดเศษขึ้นให้เป็ นทศนิยมที่ใกล้ที่สุด

คู่่�มืือคำำ�เตืือนแบบสั้้�น:การแจ้้งเตืือนระดัับพลััง

การแจ้้งเตืือนอุุ ณหภููมิิ

การแจ้้งเตืือนให้้รับั การซ่่อมบำำ�รุุง

ไฟแจ้้งเตืือน LEDเปลี่่ย� นแบตเตอรี่่�

ปล่่อยให้้เย็็นลง

ติิดต่่อฝ่่ายบริิการลููกค้้าเพื่่�อทำำ�การซ่่อม

ติิดต่่อฝ่่ายบริิการลููกค้้าเพื่่�อทำำ�การซ่่อม

• ไม่่มีีเสีียง
• กระพริิบทุุก 2 วิินาทีี
• ให้้มีีการทำำ�งาน

• บี๊๊�บติิดต่่อกััน 3 ครั้้�ง
• กระพริิบเร็็วครั้้�งเดีียว
• ห้้ามการทำำ�งาน

• แบตเตอรี่่ต่ำ� �ำ�: กระพริิบช้้า
• ปิิดเครื่่�อง: บี๊๊�บ 3 ครั้้�ง กระพริิบ
• ห้้ามการทำำ�งาน

โหมดพััก: แสงบ่่มจะเข้้าสู่่โ� หมดพัักหลัังจากไม่่มีีการใช้้งาน 60 วิินาทีี และบ่่งบอกด้้วยการกะพริิบช้้าๆของไฟ โหมด / สถานะ การยกหรืือสััมผััสตััวเครื่่�องจะเป็็นการกระตุ้้�นแสงบ่่มให้้กลัับสู่่ก� ารตั้้�งค่่าของการใช้้งานล่่าสุุดโดย
อััตโนมััติิ เพื่่�อยืืดอายุุการใช้้งานแบตเตอรี่่� ปล่่อยแสงบ่่มไว้้โดยไม่่ให้้ถููกรบกวนเมื่่�อไม่่ใช้้งาน

• บี๊๊�บ 3 ครั้้�ง
• กระพริิบช้้า
• ห้้ามการทำำ�งาน

ทำำ�ความสะอาด

1) ทิ้้�งปลอกกั้้�นในถัังขยะมาตรฐานหลัังจากใช้้กัับผู้้�ป่่วยแต่่ละราย
2) ดููส่่วนขั้้�นตอนการทำำ�งาน

คำำ�แนะนำำ�ในการติิดตั้้�งตััวยึึด

1) ควรติิดตั้้�งตััวยึึดกัับพื้้� นผิิวราบ ปราศจากน้ำำ��มันั
2) ทำำ�ความสะอาดพื้้� นผิิวด้้วยแอลกอฮอล์์สำ�ำ หรัับเช็็ด
3) ลอกด้้านหลัังเทปกาวของตััวยึึดออก
4) วางตััวยึึดในลัักษณะที่่�ให้้แสงบ่่มชี้้�ขึ้้�นเมื่่�อหยิิบออก กดให้้แน่่น

ซ่่อมบำำ�รุุง

การทำำ�ความสะอาดทั่่�วไปของเครื่่�องฉายแสง
หลัังการใช้้งานแต่่ละครั้้�ง ให้้ชุุบผ้้ากอซหรืือผ้้านุ่่�มด้้วยน้ำำ��ยาฆ่่าเชื้้�อบนพื้้� นผิิวที่่�ได้้รับั อนุุมัติั ิ แล้้วเช็็ดพื้้�นผิิวและเลนส์์ น้ำำ��ยาทำำ�ความสะอาดที่่�ไม่่ได้้รับั อนุุญาตอาจทำำ�ให้้เครื่่�องฉายแสงเสีียหายได้้
น้ำำ��ยาทำำ�ความสะอาดที่่�ยอมรัับได้้:• ไอโซโพรพิิลแอลกอฮอล์์ 70%• เอทานอล 70%280การทำำ�ความสะอาดทั่่�วไปของที่่�แผงบัังแสง:ทำำ�ความสะอาดแผงบัังแสงไร้้สาย VALO โดยใช้้น้ำ��ำ ยาฆ่่าเชื้้�อบนพื้้� นผิิว อย่่านึ่่�ง(autoclave)การซ่่อมแซม
การบำำ�รุุงรัักษาที่่�ดำำ�เนิินการโดยผู้้�ใช้้
1) ใช้้ปลอกกั้้�นเพื่่�อป้้องกัันไม่่ให้้คอมโพสิิตทางทัันตกรรมเกาะติิดกัับพื้้� นผิิวของเลนส์์ หากจำำ�เป็็น ให้้ใช้้เครื่่�องมืือทัันตกรรมที่่�เป็็นพลาสติิกหรืือสเตนเลสเพื่่�อขจััดวัสั ดุุที่่�เกาะติิดออกอย่่างระมััดระวััง อย่่าใช้้เครื่่�องมืือที่่�จะทำำ�ให้้
281

เลนส์์เสีียหาย
2) เครื่่�องวััดแสงมีีความแตกต่่างกัันอย่่างมาก และได้้รับั การออกแบบมาสำำ�หรัับหััวทิิปและเลนส์์นำำ�แสงโดยเฉพาะ Ultradent แนะนำำ�ให้้ตรวจสอบเอาต์์พุุตในโหมดพลัังงานมาตรฐานเป็็นประจำำ� หมายเหตุุ: เอาต์์พุุตตััวเลขที่่�แท้้
จริิงจะเอีียงเนื่่�องจากความไม่่ถููกต้้องของมาตรวััดแสงทั่่�วไปและชุุด LED แบบกำำ�หนดเองในไฟบ่่ม
การบำำ�รุุงรัักษาแบตเตอรี่่�
การชาร์์จและการเปลี่่ย� นแบตเตอรี่่�
ไฟบ่่มมาพร้้อมกัับแบตเตอรี่่ลิ� เิ ธีียมเหล็็กฟอสเฟตแบบชาร์์จไฟได้้ 4 ก้้อน Ultradent ขอแนะนำำ�ให้้ใช้้แบตเตอรี่่แ� บบชาร์์จไฟได้้ของ VALO เท่่านั้้�น
วิิธีีชาร์์จแบตเตอรี่่:�1) วางเครื่่�องชาร์์จและแบตเตอรี่่ใ� นบริิเวณที่่�ไม่่สัมั ผััสกัับของเหลวหรืือสเปรย์์ฆ่า่ เชื้้�อ
2) ใส่่แบตเตอรี่่ล� งในเครื่่�องชาร์์จโดยให้้ปลายขั้้�วบวก (+) ชี้้�ไปทางไฟแสดงสถานะบนเครื่่�องชาร์์จ
3) ไฟสีีเขีียวแสดงว่่าแบตเตอรี่่พร้� อ้ มใช้้งาน
4) แบตเตอรี่่จ� ะใช้้เวลาชาร์์จ 1-3 ชั่่�วโมง ทิ้้�งแบตเตอรี่่ไ� ว้้ในเครื่่�องชาร์์จจนกว่่าจะพร้้อมใช้้งาน
หมายเหตุุ: หากไฟสีีแดงบนเครื่่�องชาร์์จไม่่เปลี่่ย� นเป็็นสีีเขีียวเมื่่�อแบตเตอรี่่ช� าร์์จนานกว่่าสามชั่่�วโมง แบตเตอรี่่อ� าจเสื่่�อมสภาพ (เสีียหาย) และไม่่สามารถชาร์์จได้้ ลองแบตเตอรี่่ใ� หม่่หรืือโทรติิดต่่อฝ่่ายบริิการลููกค้้าของ
Ultradent เพื่่�อสั่่�งซื้้�อแบตเตอรี่่แ� บบชาร์์จใหม่่ได้้

ข้้อมููล CEชิ้้�น
ตััวชาร์์จ VALO

แบตเตอรี่่� VALO

ข้้อมููลทางเทคนิิค / ข้้อมููล

แถบความยาวคลื่่�นในการบ่่มคอมโพสิิตอย่่างมีีประสิิทธิิภาพ:วิิธีีเปลี่่ย� น/ใส่่แบตเตอรี่่:�1) ถอดฝาด้้านหลัังออกโดยบิิดทวนเข็็มนาฬิิกาหนึ่่�งในสี่่ร� อบ2) ถอดแบตเตอรี่่อ� อก3) ใส่่แบตเตอรี่่ใ� หม่่ด้้านบวก (+) ก่่อน
4) ใส่่ฝาปิิดด้้านหลัังกลัับเข้้าไปใหม่่โดยจััดตำำ�แหน่่งและดัันเบาๆ ขณะที่่�บิดิ ตามเข็็มนาฬิิกา ฝาปิิดจะคลิิกเมื่่�อติิดแน่่นแล้้ว
5) เครื่่�องพร้้อมใช้้งานแล้้ว

พลังแห่งแสงสเปกตรัม (มิลลวิ ตั ต์/นาโนเมตร)แถบความยาวคลื่นคอมโพสิตที่มีประสิทธภิ าพในการบ่ม
(ช่วงความยาวคลื่นที่ใช้งานได้: 385-515 นาโนเมตร)แบตเตอรี่่เ� หลืือน้้อย: ไฟแสดงการบ่่มจะส่่งสััญญาณให้้ผู้้�ใช้้ทราบว่่าถึึงเวลาเปลี่่ย� นแบตเตอรี่่เ� มื่่�อไฟแสดงสถานะแบตเตอรี่่เ� หลืือน้้อยกะพริิบเป็็นสีีแดง หากการชาร์์จแบตเตอรี่่ต่ำ� �ำ�เกิินไป เสีียงบี๊๊�บ 3 ครั้้�งจะดัังขึ้้�น และไฟบ่่มจะไม่่
อนุุญาตให้้ทำำ�งานต่่อไปจนกว่่าจะชาร์์จแบตเตอรี่่ใ� หม่่หรืือใส่่แบตเตอรี่่ใ� หม่่ (ดููคู่่�มืือคำำ�เตืือนฉบัับย่่อ)เวลาในการชาร์์จและอายุุการใช้้งานแบตเตอรี่่:� อายุุการใช้้งานแบตเตอรี่่ที่� ่�ชาร์์จจนเต็็มในไฟบ่่มจะขึ้้�นอยู่่�กัับโหมด/ช่่วงเวลา ประเภทของแบตเตอรี่่� ปริิมาณการใช้้งาน และประสิิทธิิภาพของ LED• ช่่วงเวลาการชาร์์จที่่�แนะนำำ�: ชาร์์จตามความจำำ�เป็็นหรืือเมื่่�อสััญญาณไฟแบตเตอรี่่อ่� ่อนสว่่างขึ้้�น อายุุการใช้้งานขึ้้�นอยู่่�กัับการใช้้งาน
• แบตเตอรี่่เ� สริิม: เครื่่�องฉายแสงมาพร้้อมกัับแบตเตอรี่่แ� บบชาร์์จไฟได้้ 4 ก้้อน
การซ่่อมแซมโดยผู้้�ผลิิต
1) การซ่่อมแซมจะต้้องดำำ�เนิินการโดยช่่างบริิการที่่�ได้้รับั อนุุญาตเท่่านั้้�น Ultradent จะต้้องจััดหาเอกสารประกอบให้้กัับเจ้้าหน้้าที่่�บริิการเพื่่�อทำำ�การซ่่อมแซม
2) เมื่่�อส่่งเครื่่�องไปซ่่อมแซม ซ่่อมบำำ�รุุง หรืือสอบเทีียบ ให้้ถอดแบตเตอรี่่อ� อกจากเครื่่�องฉายแสงและอุุ ปกรณ์์ชาร์์จเสมอ ห่่อแบตเตอรี่่� เครื่่�องชาร์์จ อะแดปเตอร์์ และเครื่่�องฉายแสงแยกกัันในกล่่องส่่งคืืน
3) จััดส่ง่ แบตเตอรี่่ต� ามข้้อบัังคัับท้้องถิ่่�น
การรัับประกััน
เมื่่�อใช้้งานตามคู่่�มืือการใช้้งานที่่�มาพร้้อมกัับผลิิตภััณฑ์์ เป็็นระยะเวลาหนึ่่�ง 5 ปีีนัับจากวัันที่่�ซื้�อ้ Ultradent Products, Inc. (“Ultradent”) รัับประกัันว่่าผลิิตภััณฑ์์นี้้จ� ะต้้อง (i) เป็็นไปตามข้้อกำำ�หนดในเนื้้�อหาทั้้�งหมดที่่�กำำ�หนดไว้้
ในเอกสารประกอบของ Ultradent ที่่�มาพร้้อมกัับผลิิตภััณฑ์์และ (ii) ปราศจากข้้อบกพร่่องด้้านวััสดุุและฝีีมืือการผลิิตำ ัดเฉพาะ
การรัับประกัันแบบจำำ�กััดนี้้ไ� ม่่สามารถโอนสิิทธิ์์ไ� ด้้และใช้้กัับผู้้�ซื้้�อเดิิมเท่่านั้้�น และไม่่ขยายไปยัังเจ้้าของผลิิตภััณฑ์์คนต่่อๆ ไป การรัับประกัันแบบจำำ�กััดนี้้ไ� ม่่ครอบคลุุมถึึงส่่วนประกอบอุุ ปกรณ์์เสริิมอื่่�นๆ เช่่น แต่่ไม่่จำ�กั
แบตเตอรี่่� ที่่�ชาร์์จ อะแดปเตอร์์ หรืือเลนส์์แบบปรัับได้้ การรัับประกัันแบบจำำ�กััดนี้้จ� ะถืือเป็็นโมฆะหากผลิิตภััณฑ์์ล้ม้ เหลวหรืือเสีียหายเนื่่�องจากความประมาทเลิินเล่่อ การใช้้ในทางที่่�ผิิด อุุ บัติั ิเหตุุ การดััดแปลง การปลอมแปลง การเปลี่่ย� นแปลง หรืือการไม่่ปฏิิบัติั ิตามคำำ�แนะนำำ�ในการใช้้งานที่่�เกี่่�ยวข้้อง เพื่่�อวััตถุุประสงค์์ตััวอย่่างเท่่านั้้�น ผลิิตภััณฑ์์ที่่�ตกหล่่นและเสีียหายจะไม่่ครอบคลุุมภายใต้้การรัับประกัันนี้้� เพื่่�อให้้มีีคุณุ สมบััติิภายใต้้การรัับประกัันแบบจำำ�กััดนี้้�
จะต้้องส่่งหลัักฐานการซื้้�อ (เช่่น ใบเสร็็จรัับเงิินหรืือเอกสารที่่�คล้้ายกััน) ไปยััง Ultradent พร้้อมกัับผลิิตภััณฑ์์ที่่�ชำ�รุำ ุดผลิิตภััณฑ์์ที่่�มีีข้อ้ บกพร่่องซึ่่�งตรงตามเงื่่�อนไขการรัับประกัันที่่�กำำ�หนดไว้้ในที่่�นี้้� จะต้้องได้้รับั การซ่่อมแซมหรืือเปลี่่ย� นใหม่่ตามดุุลยพิินิิจของ Ultradent แต่่เพีียงผู้้�เดีียว ไม่่ว่า่ ในกรณีีใดความรัับผิิดของ Ultradent สำำ�หรัับ
ผลิิตภััณฑ์์จะเกิินกว่่าราคาซื้้�อที่่�ผู้้�ซื้้�อชำำ�ระ ไม่่ว่า่ ในกรณีีใด Ultradent จะไม่่รับั ผิิดชอบต่่อความเสีียหายทางอ้้อม ความเสีียหายที่่�เกิิดขึ้้�นโดยบัังเอิิญ คาดการณ์์ล่ว่ งหน้้า ไม่่คาดคิิด พิิเศษหรืือเป็็นผลสืืบเนื่่�องใด ๆ ที่่�เกิิดขึ้้�นจาก
หรืือเกี่่�ยวข้้องกัับการใช้้ผลิิตภััณฑ์์นี้้�
์มีประสิทธภิ าพลดลง
แถบความยาวคลื่ น
395-415 นาโนเมตร
395-

ข้้อมููล / ข้้อมููลจำำ�เพาะ

เลนส์์

เส้้นผ่่าศููนย์์กลาง 11.7 มมช่่วงความยาวคลื่่�น

• ช่่วงความยาวคลื่่�นที่่�ใช้้ประโยชน์์ได้้: 385 - 515nm• ความยาวคลื่่�นสููงสุุด: 395 - 415nm และ 440 - 480nm
Nominal Radiant Exitance Comparison Chartการจััดเก็็บและการกำำ�จััด

เครื่่�องมืือวััด

หากจััดเก็็บแสงบ่่มนานเกิินกว่่า 2 สััปดาห์์ หรืือเก็็บไว้้สำ�ำ หรัับการเดิินทาง ให้้ถอดแบตเตอรี่่อ� อกทุุกครั้้�ง หากใส่่แบตเตอรี่่ทิ้้� �งไว้้ในเครื่่�องเป็็นระยะเวลานานโดยที่่�ไม่่ชาร์์จแบตเตอรี่่ใ� หม่่ แบตเตอรี่่อ� าจเสื่่�อมหรืือไม่่สามารถชาร์์จไฟ
ใหม่่ได้้ อย่่าเก็็บแบตเตอรี่่ไ� ว้้ในที่่�ที่่�มีีอุุณหภููมิิสููงกว่่า40 ° C (104 ° F) หรืือในที่่�ที่่�มีีแสงแดดส่่องถึึงโดยตรง
การจััดเก็็บและการขนส่่งแสงบ่่ม:• อุุ ณหภููมิิ: + 10°C ถึึง + +40°C (+ 50°F ถึึง + +104°F)• ความชื้้�นสััมพััทธ์:์ 10% ถึึง 95%• ความดัันบรรยากาศ: 500 hPa ถึึง 1,060 hPaตารางความเข้้มแสง

เมื่่�อกำำ�จัดั ขยะอิิเล็็กทรอนิิกส์์ (เช่่น แสงบ่่ม เครื่่�องชาร์์จแบตเตอรี่่� และแหล่่งจ่่ายไฟ) ให้้ปฏิิบัติั ิตามกฎระเบีียบการทิ้้�งขยะและการรีีไซเคิิลของท้้องถิ่่�น

ข้้อพิจิ ารณาด้้านเทคนิิคอุุ ปกรณ์์เสริิม

282

High Powerมาตรฐาน

คุุณลัักษณะ

ผลิิตภััณฑ์์นี้้อ� อกแบบมาเพื่่�อใช้้กัับปลอกกั้้�น ดููด้้านบนสำำ�หรัับการบำำ�รุุงรัักษาและการทำำ�ความสะอาดทั่่�วไป

แผ่่นกั้้�นแสง VALO ไร้้สาย

Xtraคลื่น อังกฤษ (นาโนเมตร)การดำำ�เนิินการ

ชิ้้�น

แถบความยาวคลื่นด้าน
บนที่มีประสิทธภิ าพ
440--480 นาโนเมตร
440ข้้อมููล CEแสงบ่่ม VALO แกรนด์์

ตััววิิเคราะห์์สเปกตรััม ‡ Gigahertzรัังสีีต่่อพื้้� นที่่�

พลัังงานทั้้�งหมด

เส้้นผ่่าศููนย์์ก-ลางของ
มิิเตอร์์

15 mm

15 mm

สแตนดาร์์ด พาวเวอร์์
(±10%)

900 mW/cm²

970 mW

High Power Plus(±10%)

1500 mW/cm²

1615 mW

พลัังงานสููงพิิเศษ
(±10%)

2100 mW/cm²

2260 mW

ระดัับ: IEC 60601-1 (ความปลอดภััย), IEC 60601-1-2(EMC)การออกจากการแผ่่รังั สีี (Radiant Exitance) จะแตก
ต่่างกัันไปขึ้้�นอยู่่�กัับความสามารถของอุุ ปกรณ์์ วิิธีีการวััด
และตำำ�แหน่่งของแสง
‡ Radiant Exitance เป็็นไปตามมาตรฐาน ISO
10650 เมื่่�อวััดด้้วยเครื่่�องวิิเคราะห์์สเปกตรััม
Gigahertz-Optikน้ำำ��หนััก:• พร้้อมแบตเตอรี่่:� 6 ออนซ์์ (190 กรััม)• ไม่่รวมแบตเตอรี่่:� 5 ออนซ์์ (150 กรััม)ขนาด: (8 x 1.28 x 1.06) นิ้้�ว (203 x 32.5 x 27) มม
283

คุุณลัักษณะ

ข้้อมููล / ข้้อมููลจำำ�เพาะ

แหล่่งจ่่ายไฟ-ของเครื่่�องชาร์์จ

ระดัับ: IEC 60601-1 (ความปลอดภััย)พลัังงานออก - 12VDC ที่่� 500mAความยาวสายไฟ - 6 ฟุุ ต (1.8 เมตร)พลัังงานเข้้า - 100VAC ถึึง 240VAC
แหล่่งจ่่ายไฟของเครื่่�องชาร์์จ VALO เป็็นแหล่่งจ่่ายไฟเกรดการแพทย์์คลาส II และให้้การ
เครื่่�องจ่่ายไฟพร้้อมปลั๊๊�กสากล Ultradent P/N 5930 VALO
แยกพลัังงานจากแหล่่งจ่่ายไฟหลััก

ตััวชาร์์จ VALO

ตััวชาร์์จแบตเตอรี่่อั� ัจริิยะ VALO 3.6VDC Lithium Iron Phosphate:• ปิิดอััตโนมััติิเมื่่�อชาร์์จเต็็ม
• ตรวจจัับแบตเตอรี่่ที่� ่�ชำ�รุำ ุดโดยอััตโนมััติิ
• การป้้องกััน: ความร้้อน, การชาร์์จมากเกิินไป, การลััดวงจร, การกลัับขั้้�ว
⸰ LED สีีแดง - กำำ�ลังั ชาร์์จ
⸰ LED สีีเขีียว - ว่่างเปล่่าหรืือชาร์์จเต็็ม
⸰ LED off - ลััดวงจร
• เวลาในการชาร์์จ: 1 - 3 ชั่่�วโม
ระดัับ: CE, WEEEแบตเตอรี่่� VALO

ชาร์์จใหม่่ได้้: เคมีีที่่�ปลอดภััย ลิิเธีียมเหล็็กฟอสเฟต (LiFePO4) RCR123A• แรงดัันใช้้งาน: 3.2VDC, 400mAH (วััตต์์ชั่่ว� โมงที่่�ต้้องการ)การให้้คะแนน: UL 62133-2, CE, RoHS, WEEEเซลล์์ได้้รับั การจััดอัันดัับสำำ�หรัับรอบการชาร์์จ/คายประจุุ >2000 รอบก่่อนที่่�จะถึึง 70% ของความจุุ แนะนำำ�ให้้เปลี่่ย� นแบตเตอรี่่ทุ� กุ ๆ 3-5 ปีี แต่่อายุุการใช้้
งานอาจแตกต่่างกัันไปขึ้้�นอยู่่�กัับการใช้้งาน เปลี่่ย� นใหม่่ตามความจำำ�เป็็น

สภาวะการใช้้งานอุุ ณหภููมิิ: + 10°C ถึึง + 32°C (+ 50°F ถึึง + 90°F)ความชื้้�นสััมพััทธ์:์ 10% ถึึง 95%ความดัันบรรยากาศ: 700 hPa ถึึง 1060 hPaรอบการทำำ�งาน:แสงบ่่มถููกออกแบบมาสำำ�หรัับการทำำ�งานระยะสั้้�น ที่่�อุุณหภููมิิสููงสุุด (32 ° C) หลัังจากที่่�ใช้้ไป 1 นาทีี แบบไม่่ได้้หยุุดพักั ให้้ปิดิ 30 นาทีี (ระยะเวลาระบาย
ความร้้อน)การแจ้้งปััญหา
หากการแก้้ปััญหาที่่�ได้้แนะนำำ�ไว้้ด้้านล่่างนี้้�ไม่่สามารถแก้้ไขปััญหาได้้ กรุุณาโทรติิดต่่อ Ultradent ที่่�หมายเลข 800.552.5512 นอกสหรััฐอเมริิกาโปรดติิดต่่อผู้้�จััดจำ�ำ หน่่าย Ultradent หรืือตััวแทนจำำ�หน่่ายอุุ ปกรณ์์ทัันตกรรม
ปััญหา

การแก้้ไขปััญหาที่่�เป็็นไปได้้

แสงไม่่เปิิดขึ้้�น

1) กดปุ่่�ม เวลา / โหมด หรืือ ปุ่่�มพาวเวอร์์ เพื่่�อปลุุกจากโหมดประหยััดพลังั งาน2) ตรวจสอบไฟแสดงสถานะแบตเตอรี่่ต่ำ� �ำ�สีีแดงเพื่่�อดููสถานะการชาร์์จแบตเตอรี่่�
3) ตรวจสอบว่่าใส่่แบตเตอรี่่ใ� หม่่เข้้าไปในเครื่่�องอย่่างถููกต้้องหรืือไม่่
4) หากไฟเตืือน LED เป็็นสีีแดงและสีีเหลืืองกระพริิบหมายความว่่าแสงบ่่มถึึงขีีดจำำ�กััดอุุณหภููมิิภายในที่่�ปลอดภััยแล้้ว ปล่่อยให้้แสงบ่่มให้้เย็็นลงเป็็น
เวลา 10 นาทีีหรืือใช้้ผ้้าเย็็นและชื้้�นเช็็ดเพื่่�อทำำ�ให้้ตััวเครื่่�องเย็็นลงอย่่างรวดเร็็ว
5) หากไฟเตืือน LED สีีแดงกะพริิบและส่่งเสีียงบี๊๊�บอย่่างต่่อเนื่่�องให้้ติิดต่่อฝ่่ายบริิการลููกค้้า Ultradent เพื่่�อทำำ�การซ่่อมแซม

ไฟไม่่สว่่างในระยะเวลาที่่�ต้้องการ

1) ตรวจสอบไฟโหมดและการกำำ�หนดระยะเวลา เพื่่�อการป้้อนเวลาที่่�ถููกต้้อง
2) ตรวจสอบไฟสถานะแบตเตอรี่่ต่ำ� �ำ�เพื่่�อดููสถานะการชาร์์จแบตเตอรี่่�
3) ตรวจสอบว่่าใส่่แบตเตอรี่่ใ� หม่่เข้้าไปในเครื่่�องอย่่างถููกต้้องหรืือไม่่

แสงไม่่บ่ม่ เรซิินอย่่างถููกต้้อง

1) ตรวจสอบเลนส์์เพื่่�อหาเรซิิน / คอมโพสิิตที่่�ตกค้้าง
2) สวมแว่่นสีีอำำ�พันั เพื่่�อปกป้้องดวงตาจากรัังสีี UV, ตรวจสอบว่่าแสง LED ทำำ�งานอยู่่�
3) ตรวจระดัับพลัังงานด้้วยมิิเตอร์์วัดั แสง หากใช้้เครื่่�องวััดแสง Ultradentบัันทึึก: ตััวเลขบอกพลัังแสงที่่�ปล่่อยออกมาตามจริิงจะถููกบิิดเบืือนไป เนื่่�องจากความคลาดเคลื่่�อนของตััวตรวจวััดแสงทั่่�วไป และชุุดไฟ LED พิิเศษที่่�ใช้้
มิิเตอร์์วัดั แสงมีีความหลากหลายมาก และได้้รับั การออกแบบมาสำำ�หรัับปลายนำำ�แสงและเลนส์์ เฉพาะรุ่่�น
4) ตรวจสอบวัันหมดอายุุของเรซิ่่�นสำำ�หรัับบ่่ม
5) ตรวจสอบให้้แน่่ใจว่่ามีีการปฏิิบัติั ิตามเทคนิิคที่่�เหมาะสมตามคำำ�แนะนำำ�ของผู้้�ผลิิต

แบตเตอรี่่ไ� ม่่ชาร์์จ

1) ตรวจสอบให้้แน่่ใจว่่าใส่่แบตเตอรี่่ล� งในเครื่่�องชาร์์จในทิิศทางที่่�ถููกต้้องและให้้แบตเตอรี่่ช� าร์์จเป็็นเวลา 1-3 ชั่่�วโมง
2) หากไฟสีีแดงบนตััวชาร์์จไม่่เปลี่่ย� นเป็็นสีีเขีียว ให้้โทรติิดต่่อฝ่่ายบริิการลููกค้้า Ultradent เพื่่�อสั่่�งแบตเตอรี่่แ� ละ / หรืืออุุ ปกรณ์์ชาร์์จทดแทน
3) หากมองไม่่เห็็นไฟสีีเขีียวหรืือสีีแดงบนตััวชาร์์จให้้โทรติิดต่่อฝ่่ายบริิการลููกค้้า Ultradent เพื่่�อสั่่�งซื้้�อหรืือเปลี่่ย� นอุุ ปกรณ์์ชาร์์จและ / หรืืออะแดป
เตอร์์ ACตััวชาร์์จไม่่ชาร์์จแบตเตอรี่่�

1) ตรวจสอบให้้แน่่ใจว่่าเสีียบปลั๊๊�กเครื่่�องชาร์์จแล้้วและเสีียบอะแดปเตอร์์ AC เข้้ากัับปลั๊๊�กไฟที่่�ใช้้งานได้้
2) หากมองไม่่เห็็นไฟสีีเขีียวหรืือสีีแดงบนอุุ ปกรณ์์ชาร์์จ ให้้โทรติิดต่่อฝ่่ายบริิการลููกค้้า Ultradent สำำ�หรัับอุุ ปกรณ์์ชาร์์จและ / หรืืออะแดปเตอร์์ ACใหม่่ไม่่สามารถเปลี่่ย� นโหมดหรืือช่่วงเวลา

1) กดทั้้�งปุ่่�มเวลา / โหมดและปุ่่�มเพาเวอร์์ค้้างไว้้จนกว่่าสััญญาณเสีียงบี๊๊�บจะแจ้้งเตืือนว่่าแสงบ่่มได้้ถููกปลดล็็อคแล้้ว

ข้้อมููลเบ็็ดเตล็็ด
คำำ�แนะนำำ�และประกาศเรื่่�องการปล่่อยคลื่่�นแม่่เหล็็กไฟฟ้้า
แสงบ่่มมีีไว้้สำ�ำ หรัับใช้้ในสภาพแวดล้้อมแม่่เหล็็กไฟฟ้้าที่่�ระบุุด้้านล่่าง ลููกค้้าหรืือผู้้�ใช้้ควรทำำ�ให้้แน่่ใจว่่ามีีการใช้้งานในสภาพแวดล้้อมดัังกล่่าว
การทดสอบการปล่่อยปล่่อย
คลื่่�นแม่่เหล็็กไฟฟ้้า

284

การปฏิิบัติั ิตาม

สภาพแวดล้้อมทางคลื่่�นแม่่เหล็็กไฟฟ้้า - คำำ�แนะนำำ�

การปล่่อยคลื่่�นความถี่่�วิทยุิ ุ
CISPR 11=กลุ่่�มที่่� 1แสงบ่่มใช้้แบตเตอรี่่แ� ละจะไม่่ได้้รับั ผลกระทบจาก EMI, RF หรืือการป้้องกัันไฟกระชาก

การปล่่อยคลื่่�นความถี่่�วิทยุิ ุ
CISPR 11คลาส Bการปล่่อยฮาร์์มอนิิก
IEC 61000-3-2

N/Aการปล่่อยแรงดัันกระเพื่่�อม / กระพริิบ
IEC 61000-3-

N/Aแสงบ่่มใช้้พลังั งานไฟฟ้้าและแม่่เหล็็กไฟฟ้้าสำำ�หรัับการทำำ�งานภายในเท่่านั้้�น ดัังนั้้�น การปล่่อยคลื่่�นความถี่่�วิทยุิ ุใดๆ จึึงต่ำำ��มาก และไม่่น่า่ จะก่่อให้้เกิิดการรบกวนอุุ ปกรณ์์อิิเล็็กทรอนิิกส์์ในบริิเวณใกล้้เคีียง
การทดสอบการปล่่อยฮาร์์มอนิิกและความผัันผวนของแรงดัันไฟฟ้้าใช้้ไม่่ได้้กัับแสงบ่่มเนื่่�องจากแสงบ่่มใช้้พลังั งานจากแบตเตอรี่่�
แสงบ่่มเหมาะสำำ�หรัับการใช้้งานในทุุกสถานประกอบการ รวมถึึงสถานประกอบการในประเทศและสถานที่่�ที่่�เชื่่�อมต่่อ
โดยตรงกัับเครืือข่่ายแหล่่งจ่่ายไฟแรงดัันต่ำำ��สาธารณะ

285

คำำ�แนะนำำ�และประกาศเกี่่�ยวกัับภููมิิคุ้้�มกัันแม่่เหล็็กไฟฟ้้า

คำำ�แนะนำำ�และประกาศเกี่่�ยวกัับภููมิิคุ้้�มกัันแม่่เหล็็กไฟฟ้้าสำำ�หรัับระบบที่่�ไม่่ใช่่ระบบสนัับสนุุนการยัังชีีพ

แสงบ่่มมีีไว้้สำ�ำ หรัับใช้้ในสภาพแวดล้้อมแม่่เหล็็กไฟฟ้้าที่่�ระบุุด้้านล่่าง ลููกค้้าหรืือผู้้�ใช้้ควรทำำ�ให้้แน่่ใจว่่ามีีการใช้้งานในสภาพแวดล้้อมดัังกล่่าว
การทดสอบภููมิิคุ้้�มกััน

ระดัับการทดสอบ IEC 60601

ระดัับการปฏิิบัติั ิตามกฎระเบีียบ

คำำ�แนะนำำ�เกี่่�ยวกัับสภาพแวดล้้อมทางแม่่เหล็็กไฟฟ้้า

การถ่่ายเทไฟฟ้้าสถิิต (ESD)ติิดต่่อ± 8 kVอากาศ± 15 kVติิดต่่อ± 8 kVอากาศ± 15 kVควรจำำ�กััดสภาพแวดล้้อมทางกายภาพต่่อไปนี้้�:1) รหััส IP: IP202) ห้้ามแช่่ในของเหลว
3) ห้้ามใช้้ใกล้้กัับแก๊๊สไวไฟ อุุ ปกรณ์์เป็็นแบบ Non-APG
และ Non-AP4) ช่่วงความชื้้�นในการจััดเก็็บ: 10% - 95%5) ช่่วงอุุ ณหภููมิิในการจััดเก็็บ: 10° C - 40° C

IEC 61000-4-2การเกิิดแรงดัันไฟฟ้้าเกิินชั่่�วครู่่แ� บบรวดเร็็ว
IEC 61000-4-4แรงดัันสำำ�หรัับสายแหล่่งจ่่ายไฟ ± 2 kVแรงดัันสำำ�หรัับสายอิินพุุ ต / สายเอาต์์พุุต ± 1 kVแรงดัันสำำ�หรัับสายแหล่่งจ่่ายไฟ ± 2 kVหมายเหตุุ 1: แสงบ่่มไม่่มีีพอร์์ต I / 0แสงบ่่มใช้้พลังั งานจากแบตเตอรี่่แ� ละไม่่สามารถเชื่่�อมต่่อกัับแห
ลงจ่่ายไฟ AC ได้้ไฟฟ้้ากระชาก
IEC 61000-4-5แรงดัันระหว่่างสายไฟ ± 1 kVแรงดัันระหว่่างสายไฟและสายดิิน ± 2 kVแรงดัันระหว่่างสายไฟ ± 1 kVแรงดัันระหว่่างสายไฟและสายดิิน ± 2 kVเนื่่�องจากแสงบ่่มใช้้พลังั งานจากแบตเตอรี่่� จึึงไม่่ได้้รับั ผลกระ
ทบจากกระแสไฟฟ้้าเปลี่่ย� นสภาพชั่่�วครู่่ � ไฟกระชาก แรงดัันไฟฟ้้า
ลดชั่่�วครู่่ � ไฟไม่่พอ การรบกวนหรืือการเปลี่่ย� นแปลงพลัังงาน
บน AC MAINSแรงดัันไฟฟ้้าตกชั่่�วขณะ การลััดวงจร การ <5% Uหยุุดชะงััก และการเปลี่่ย� นแปลงกัับสาย
(แรงดัันไฟฟ้้าตกชั่่�วขณะ >95% ใน Uไฟจ่่ายพลัังงาน
สำำ�หรัับ 0.5 รอบ)<5% U(แรงดัันไฟฟ้้าตกชั่่�วขณะ >95% ใน Uสำำ�หรัับ 0.5 รอบ)40% U(แรงดัันไฟฟ้้าตกชั่่�วขณะ 60% ใน Uสำำ�หรัับ 5 รอบ)40% U(แรงดัันไฟฟ้้าตกชั่่�วขณะ 60% ใน Uสำำ�หรัับ 5 รอบ)70% U(แรงดัันไฟฟ้้าตกชั่่�วขณะ 30% ใน Uสำำ�หรัับ 25 รอบ)70% U(แรงดัันไฟฟ้้าตกชั่่�วขณะ 30% ใน Uสำำ�หรัับ 25 รอบ)<5% U(แรงดัันไฟฟ้้าตกชั่่�วขณะ >95% ใน Uเป็็นเวลา 5 วิินาทีี)<5% U(แรงดัันไฟฟ้้าตกชั่่�วขณะ >95% ใน Uเป็็นเวลา 5 วิินาทีี)หมายเหตุุ 2: การกู้้�คืืนตนเอง

30 A/m30 A/mIEC 61000-4-11

ความถี่่�พลังั งานไฟฟ้้า
(50/60 เฮิิรตซ์์) สนามแม่่เหล็็ก
IEC 61000-4-8เครื่่�องชาร์์จแบตเตอรี่่เ� สริิมอาจได้้รับั ผลกระทบจากปรากฎการณ์์
ด้้านบน แต่่มันั เป็็นหน่่วยแยกจากกัันและไม่่ส่ง่ ผลรุุนแรงต่่อการ
ทำำ�งานของแสงบ่่มไร้้สาย VALO
หากแรงดัันแบตเตอรี่่ข� องแสงบ่่มลดลงถึึง 4VDC เครื่่�องจะไม่่
ทำำ�งาน แสงบ่่มไร้้สาย VALO จะดัับลง เมื่่�อใส่่แบตเตอรี่่ใ� หม่่และ
พลัังงานกลัับคืืนไปที่่�ระดัับที่่�เหมาะสม แสงบ่่มจะรีีสตาร์์ทและ
กลัับสู่่ส� ถานะเดิิมก่่อนที่่�จะสููญเสีียพลัังงาน แสงบ่่มจะกู้้�คืืนตััว
เองในกรณีีที่่�ไฟฟ้้าดัับ

สนามแม่่เหล็็กความถี่่�กำำ�ลังั ควรอยู่่ใ� นระดัับที่่�พบในอาคารทั่่�วไป ในที่่�อยู่่อ� าศััย สถานที่่�ดููแลสุุขภาพภายในบ้้าน อาคารพาณิิชย์์ โรงพยาบาล หรืือสภาพแวดล้้อมทางการทหา

การนำำ�คลื่่�อนความถี่่�วิทยุิ ุ

3 Vrms3 VrmsIEC 61000-4-6

150 kHz ถึึง 80 MHz

150 kHz ถึึง 80 MHzคลื่่�นวิิทยุุ RF ที่่�ปล่่อยออก

3 V/m

3 V/m

ไม่่ควรใช้้อุุปกรณ์์สื่�อ่ สาร RF แบบพกพาและมืือถืือใกล้้กัับส่่วนใดส่่วนหนึ่่�งของ VALO และสายเคเบิิลเกิินระยะ
ห่่างที่่�คำำ�นวนจากสมการที่่�สัมั พัันธ์์กัับความถี่่�ของเครื่่�องส่่งสััญญาณ
การแยกระยะห่่างที่่�แนะนำำ�:80 MHz ถึึง 800 MHz

800 MHz ถึึง 2.5 GHz
P คืือค่่าพลัังงานส่่งออกสููงสุุดของเครื่่�องส่่งสััญญาณมีีหน่่วยเป็็นวััตต์์ (W) ตามผู้้�ผลิิตเครื่่�องส่่งสััญญาณ
และ d คืือระยะห่่างที่่�แนะนำำ�มีีหน่่วยเป็็นเมตร (m)ความเข้้มของสนามจากเครื่่�องส่่งสััญญาณ RF แบบอยู่่�กัับที่่�ซึ่่�งกำำ�หนดโดยการสำำ�รวจสนามแม่่เหล็็กไฟฟ้้าaควรน้้อยกว่่าระดัับที่่�กำำ�หนดในแต่่ละช่่วงความถี่่�b

IEC 61000-4-3

80 MHz ถึึง 2.5 GHz

80 MHz ถึึง 2.5 GHzอาจมีีการรบกวนเกิิดขึ้้�นในบริิเวณใกล้้เคีียงกัับอุุ ปกรณ์์ที่่�มีีสัญั ลัักษณ์์ต่่อไปนี้้�:หมายเหตุุ 1 ณ ย่่านความถี่่� 80 MHz และ 800 MHz จะใช้้ช่ว่ งความถี่่�ที่่�สููงขึ้้�น
หมายเหตุุ 2 อาจใช้้แนวทางเหล่่านี้้�ไม่่ได้้ในทุุกสถานการณ์์ การแผ่่รังั สีีแม่่เหล็็กไฟฟ้้าได้้รับั ผลกระทบจากการดููดซัับและการสะท้้อนจากโครงสร้้าง สิ่่�งของ และผู้้�คน
a ความแรงของสนามจากเครื่่�องส่่งสััญญาณแบบอยู่่�กัับที่่� เช่่น สถานีีฐานสััญญาณวิิทยุุ (เซลลููล่่าร์์ / ไร้้สาย) โทรศััพท์์และวิิทยุุมืือถืือทางบก วิิทยุุสมััครเล่่น การออกอากาศวิิทยุุ AM และ FM และการ
ออกอากาศทางโทรทััศน์์ไม่่สามารถคาดการณ์์ได้้อย่่างแม่่นยำำ�ในทางทฤษฎีี ในการประเมิินสภาพแวดล้้อมทางคลื่่�นแม่่เหล็็กไฟฟ้้าอัันเนื่่�องมาจากเครื่่�องส่่งสััญญาณคลื่่�นวิิทยุุแบบอยู่่�กัับที่่� ควรพิิจารณา
การสำำ�รวจพื้้� นที่่�แม่่เหล็็กไฟฟ้้า หากความเข้้มของสนามที่่�วัดั ได้้ในตำำ�แหน่่งที่่�ใช้้แสงบ่่มเกิินกว่่าระดัับที่่�กำำ�หนดสำำ�หรัับ RF ที่่�ใช้้บังั คัับข้้างต้้น ควรสัังเกตแสงบ่่มไร้้สาย VALO เพื่่�อตรวจสอบการทำำ�งานว่่า
ปกติิดีีหรืือไม่่ หากสัังเกตเห็็นความผิิดปกติิในประสิิทธิิภาพการทำำ�งาน อาจจำำ�เป็็นต้้องมีีมาตรการเพิ่่�มเติิมเช่่น การจััดวางใหม่่หรืือเปลี่่ย� นตำำ�แหน่่งแสงบ่่มbในช่่วงความถี่่� 150 kHz ถึึง 80 MHz ความเข้้มของสนามไฟฟ้้าควรน้้อยกว่่า 3 V / mคำำ�แนะนำำ�และคำำ�แถลงของผู้้�ผลิิตสำำ�หรัับการแยกระยะห่่างที่่�แนะนำำ�ระหว่่างอุุ ปกรณ์์สื่�อ่ สาร RF แบบพกพาและแบบเคลื่่�อนที่่�แสงกัับบ่่มไร้้สาย VALO
แสงบ่่มมีีไว้้สำ�ำ หรัับใช้้ในสภาพแวดล้้อมแม่่เหล็็กไฟฟ้้าที่่�ควบคุุมการการรบกวนของคลื่่�น RF ผู้้�ใช้้แสงบ่่มสามารถช่่วยป้้องกัันการรบกวนทางแม่่เหล็็กไฟฟ้้าโดยรัักษาระยะห่่างขั้้�นต่ำำ��ระหว่่างอุุ ปกรณ์์สื่�อ่ สารRF แบบพกพาและแบบเคลื่่�อนที่่� (เครื่่�องส่่งสััญญาณ) และไฟบ่่มตามที่่�แนะนำำ�ด้้านล่่างตามกำำ�ลังั ส่่งออกสููงสุุดของอุุ ปกรณ์์สื่�อ่ สารหมายเหตุุ 1: แสงบ่่มไม่่ได้้ติิดตั้้�งพอร์์ตใด ๆ หรืือสาย I / O ที่่�สามารถเข้้าถึึงได้้

ระยะห่่างที่่�ถืือตามความถี่่�ของเครื่่�องส่่งสััญญาณ
(เมตร)

คำำ�แนะนำำ�และประกาศเกี่่�ยวกัับภููมิิคุ้้�มกัันแม่่เหล็็กไฟฟ้้าสำำ�หรัับระบบที่่�ไม่่ใช่่ระบบสนัับสนุุนการยัังชีีพ

กำำ�ลังั ไฟฟ้้าส่่งออกสููงสุุดที่่�กำำ�หนด
ของเครื่่�องส่่งสััญญาณ
(P ในหน่่วยวััตต์์)150 kHz – 80 MHz

80 MHz – 800 MHz

800 MHz – 2.5 GHz

0.01

0.12 เมตร

0.035 เมตร0.07 เมตร

0.1

0.37 เมตร

0.11 เมตร

0.22 เมตร

1

1.7 เมตร

0.35 เมตร

0.7 เมตร

10

3.7 เมตร

1.11 เมตร

2.22 เมตร

100

11.7 เมตร

3.5 เมตร

7.0 เมตร

แสงบ่่มมีีไว้้สำ�ำ หรัับใช้้ในสภาพแวดล้้อมแม่่เหล็็กไฟฟ้้าที่่�ระบุุด้้านล่่าง ลููกค้้าหรืือผู้้�ใช้้ควรทำำ�ให้้แน่่ใจว่่ามีีการใช้้งานในสภาพแวดล้้อมดัังกล่่าว
การทดสอบภููมิิคุ้้�มกััน

ระดัับการทดสอบ IEC
60601

ระดัับการปฏิิบัติั ิตาม
กฎ-ระเบีียบ

คำำ�แนะนำำ�เกี่่�ยวกัับสภาพแวดล้้อมทางแม่่เหล็็กไฟฟ้้า

แสงบ่่มได้้รับั การทดสอบตามมาตรฐาน IEC 60601-1-2: 2014 และผ่่านการทดสอบภายใต้้ความเข้้มของสนามที่่� 10 V / m ระหว่่าง 80-MHz ถึึง 2.5 GHz ค่่าของ 3Vrms สอดคล้้องกัับ V1 และค่่า 10 V / m สอดคล้้องกัับ E1 ในสููตรด้้านบน
สำำ�หรัับเครื่่�องส่่งสััญญาณที่่�กำำ�หนดกำำ�ลังั ไฟฟ้้าส่่งออกสููงสุุดที่่�ได้้แสดงไว้้ด้้านบน สามารถประมาณระยะห่่างที่่�แนะนำำ� d ในหน่่วยเมตร (m) ได้้โดยใช้้สมการที่่�ใช้้กัับความถี่่�ของเครื่่�องส่่งสััญญาณได้้
โดยที่่� P เป็็นกำำ�ลังั ไฟฟ้้าส่่งออกสููงสุุดของเครื่่�องส่่งสััญญาณในหน่่วยวััตต์์ ( W) โดยถืือตามผู้้�ผลิิตเครื่่�องส่่งสััญญาณ
หมายเหตุุ 1 ที่่� 80 MHz และ 800 MHz ต้้องคำำ�นึึงถึึงการแยกระยะห่่างสำำ�หรัับช่่วงความถี่่�ที่่�สููงกว่่า
หมายเหตุุ 2 อาจใช้้แนวทางเหล่่านี้้�ไม่่ได้้ในทุุกสถานการณ์์ การแผ่่รังั สีีแม่่เหล็็กไฟฟ้้าได้้รับั ผลกระทบจากการดููดซัับและการสะท้้อนจากโครงสร้้าง สิ่่�งของ และผู้้�คน
286

287

VALO™ GRAND

TRIŞIKLI DOLGU CİHAZI

Product DescriptionGeniş bant spektrumu ile VALO Grand ışık cihazı, ISO 10650 uyarınca 385-515nm dalga boyu aralığında ışıkla kürlenen tüm ürünleri polimerize etmek için tasarlanmıştır. VALO Grandışık cihazı, Ultradent VALO şarj edilebilir pilleri ve pil şarj cihazını kullanır. Işık cihazı, standart bir diş ünitesi braketine dayanacak şekilde tasarlanmıştır veya kit ile birlikte verilen braket
kullanılarak özel olarak monte edilebilir.

VALO Grand Ürün Bileşenleri:
•
•
•
•
•
•

1) Işık cihazını kullanmadan önce pilleri şarj edin (Pil Bakımı bölümüne bakın).
2) Her kullanımdan önce, ışık cihazının üzerine yeni bir bariyer kılıfı yerleştirin ve en iyi sonuçları elde etmek için lensin üzerindeki kırışıklıkları en aza indirin.
Hijyenik bariyer kılıfı ışıklı dolgu cihazına özel olarak tasarlanmıştır ve cihazının yüzeyini temiz tutar. Bariyer kılıf çapraz kontaminasyonu, dental kompozit materyallerin lens ve çubuk
gövdesinin yüzeyine yapışmasını ve temizleme solüsyonlarından kaynaklanan renk bozukluğunun ve korozyonu engellemeye yardımcı olur.

Not:• Çapraz kontaminasyonu önlemeye ve dental kompozit malzemenin lens ve çubuk gövdesinin yüzeyine yapışmasını önlemeye yardımcı olmak için, her kullanımda VALO ışık cihazınınüzerinde Ultradent onaylı bir bariyer kılıfı kullanılmalıdır. Bariyer kılıfları tek hastada kullanım için tasarlanmıştır.

VALO Kablosuz Işık Kalkanı:

VALO Kablosuz Işık Kalkanı oval şekillidir ve maksimum kullanım için döndürülebilir. Ayrıca şeffaf bir bariyer kılıfı ile kullanılabilir.

FotoğrafDüğmeler ve Işıklar

Semboller

Güç Düğmeleri(Üst veya Alt)
Zamanlama IşıklarıDüşük Pil Seviyesi – Gösterge
Mod/Durum Işığı
Zamanlama/Mod DeğiştirmeDüğmesi(Alt)Adım Adım Talimatlar
Hazırlık

Hijyenik Bariyer Kılıflarını Takma:

1 – VALO Grand ışık cihazı
4 – Ultradent VALO şarj edilebilir piller
1 – Tıbbi sınıf 12VDC AC güç adaptörlü Ultradent VALO pil şarj cihazı
1 – VALO Bariyer Kılıfı numune paketi
1 – VALO Grand ışık kalkanı
1 – Çift yapışkan bantlı ışık cihazı yüzey montaj braketi

Kontroller Hakkında Genel Bilgi:• Ekipmana zarar vermekten kaçınmak için, altın kontakları veya pil bölmesinin herhangi bir parçasını temizlemeye ÇALIŞMAYIN. Bir endişeniz varsa Ultradent Müşteri Hizmetleri'ni
arayın.
• Çapraz kontaminasyon riskini önlemek için bariyer kılıfları tek hastada kullanılır.
• Korozyon riskini azaltmak için, kullanımdan sonra bariyer kılıfını çıkarın.
• Reçinelerin yetersiz kürlenme riskini azaltmak için, lens hasarlıysa ışık cihazını kullanmayın.

Kullanım

1) Her bir güç modu, fotobaşlatıcılar ile dental materyallerin polimerizasyonu için kullanılır. Tavsiye edilen polimerizasyon süreleri için Hızlı Mod Kılavuzuna bakın.
NOT: Işıklı dolgu cihazı sırasıyla Standart Güç – Yüksek Güç Plus – Ekstra Güç moduna geçecek şekilde programlanmıştır. Örneğin, Standart Güç modundan Ekstra Güç moduna geçmek
için önce Yüksek Güç Plus moduna, ardından Ekstra Güç moduna geçmek gerekir.
2) Işıklı dolgu cihazı her modda son kullanılan zaman aralığını kaydeder ve mod değiştirildiğinde veya cihazın pilleri çıkartıldığında o zaman aralığına döner.

ÇalıştırmaPOLİMERİZASYON MODU: Standart Güç modu

ZAMAN ARALIKLARI: 5, 10, 15, ve 20 saniye.
• Işıklı dolgu cihazı İLK açıldığında varsayılan olarak bu moda ayarlıdır. Mod/Durum Işığı yeşil yanarken dört adet yeşil Zamanlama Işığı yanar, bu durum cihazın Standart Güçmodunda olduğunu gösterir.
• Zaman aralıklarını değiştirmek için Süre/Mod Düğmesine hızlı basın.
• Polimerizasyon için Güç Düğmesine basın. Zaman aralığı tamamlanmadan polimerizasyonu durdurmak için tekrar Güç Düğmesine basın.

POLİMERİZASYON MODU: Yüksek Güç Plus modu

(Üst)

Açıklaması yapılan ürünleri kullanmadan önce bütün talimatları ve GBF'yi dikkatli bir şekilde okuyun ve anlayın.

Kullanım Endikasyonları/Kullanım AmacıIşıkla aktive olan dental restoratif materyallerin ve yapışkanların polimerizasyonu için ışık kaynağı.

Uyarılar ve Önlemler
Risk Grubu 2
UYARI Bu üründen UV ışınları yayılmaktadır. Maruz kalınması durumunda gözde veya ciltte tahriş oluşabilir. Uygun bir koruyucu kullanın.
DİKKAT Bu üründen tehlikeli olması muhtemel optik radyasyon yayılmaktadır. Cihazdan çıkan ışığa doğrudan bakmayın. Gözlere zarar verebilir.
• Doğrudan ışık çıkışına BAKMAYIN. Işık cihazı kullanılırken hasta, klinisyen ve asistanlar her zaman amber renkli UV göz koruması takmalıdır.
• Elektrik çarpması riskini önlemek için bu ekipmanda değişiklik yapılmasına izin verilmez. Yalnızca birlikte verilen Ultradent VALO güç kaynağını ve fiş adaptörlerini kullanın. Bubileşenler hasar görmüşse, değiştirmeyin ve değiştirme siparişi vermek için Ultradent Müşteri Hizmetleri'ni arayın.
• Taşınabilir RF iletişim ekipmanı 30 cm'den (12 inç) daha yakın kullanılırsa performansı düşürebilir.
• Yanlış çalışmayı, artan elektromanyetik emisyonları veya azalan elektromanyetik bağışıklığı önlemek için yalnızca yetkili aksesuarları, kabloları ve güç kaynaklarını kullanın(Elektromanyetik Emisyonlar bölümüne bakın).
• Pillerin taşınmasıyla ilişkili elektrik yangını riskini önlemek için:⸰ Şarj edilebilir pilleri, şarj edilemeyen pillerle veya diğer pil türleriyle KARIŞTIRMAYIN.
⸰ Şarj edilemeyen pilleri şarj etmeye ÇALIŞMAYIN. Şarj edilemeyen pillerin şarj edilmesi yaralanmaya veya maddi hasara neden olabilir.
⸰ Pilleri, pil kontaklarını, şarj cihazını veya AC güç adaptörünü herhangi bir sıvıyla otoklavlamayın veya püskürtmeyin. Pil şarj cihazının temas noktalarında korozyon görülürse,yenisini sipariş etmek için Ultradent Müşteri Hizmetlerini arayın.
⸰ Pilleri yanıcı malzemelerin etrafında şarj ETMEYİN.
⸰ Şarj cihazını klinik işletimde TUTMAYIN.Yaralanma riskini önlemek için, korozyona uğramış (paslanmış), ezilmiş, koku veya sıvı yayan, ambalajı yırtılmış veya eksik olan veya başka
bir şekilde hasar görmüş pilleri KULLANMAYIN. Yedek pil sipariş etmek için Ultradent Müşteri Hizmetlerini arayın.
• Yalnızca önerilen pilleri kullanın. Alternatif piller arızaya neden olabilir.
• Termal tahriş veya yaralanma riskini önlemek için arka arkaya kürleme döngülerinden kaçının ve oral yumuşak dokuları herhangi bir modda 10 saniyeden fazla yakın mesafeye maruz
bırakmayın. Daha uzun kürlenme süreleri gerekiyorsa, döngüler arasında dinlenme süreleri olan çoklu kürleme döngüleri kullanın veya yumuşak dokunun ısınmasını önlemek için çiftkürlü bir ürün kullanın.
• Olumsuz fotobiyolojik reaksiyonlardan veya hassasiyetlerden muzdarip hastaları, kemoterapi tedavisi gören hastaları veya ışığa duyarlı ilaçlarla tedavi edilen hastaları tedavi ederken
dikkatli olun.
• Bu ünite, programlamayı bozabilecek güçlü manyetik veya statik elektrik alanlarına duyarlı olabilir. Bunun oluştuğundan şüpheleniyorsanız, üniteyi anlık olarak çıkarın ve prize
yeniden takın.
• Işık cihazını kostik veya aşındırıcı temizleyicilerle silmeyin, otoklavlamayın veya herhangi bir ultrasonik banyoya, dezenfektana, temizleme solüsyonuna veya sıvıya batırmayın. Dahil
edilen işleme talimatlarına uyulmaması, ışık cihazını çalışmaz hale getirebilir.
• Ekipmanın zarar görmesini önlemek için, ışık cihazının pil bölmesine parmakları, aletleri veya diğer nesneleri SOKMAYIN.
288

ZAMAN ARALIKLARI: 1, 2, 3, ve 4 saniye.
• Standart Güç modunda iken Süre/Mod Değiştirme Düğmesine 2 saniye basılı tutun ve bırakın. Mod/Durum Işığı turuncu yanarken dört adet yeşil Zamanlama Işığı yanıp söner, bu
durum cihazın Yüksek Güç modunda olduğunu gösterir.
• Zaman aralıklarını değiştirmek için Süre/Mod Düğmesine hızlı basın.
• Polimerizasyon için Güç Düğmelerinden birine basın. Bir zamanlama aralığının tamamlanmasından önce polimerizasyonu durdurmak için, Güç Düğmelerinden birine tekrar basın.
• Standart Güç moduna dönmek için Süre/Mod Değiştirme Düğmesine 2 saniye basılı tutup bırakın, böylece Ekstra Güç moduna geçilir. 2 saniye boyunca tekrar basılı tutun ve bırakın.
Mod/Durum Işığı yeşil yanarken dört adet yeşil Zamanlama Işığı yanar, bu durum cihazın Standart modda olduğunu gösterir.

POLİMERİZASYON MODU: Ekstra Güç modu

ZAMAN ARALIĞI: Sadece 3 saniye (Not: Ekstra Güç modunda, art arda polimerizasyon sırasında ısıyı sınırlandırmak için her polimerizasyon döngüsünden sonra 2 saniye güvenlik
gecikmesi bulunmaktadır. Gecikmenin sonunda, bip sesi ünitenin kullanıma hazır olduğunu gösterir).
• Standart Güç modunda iken, Süre/Mod Değiştirme Düğmesine 2 saniye basın, bırakın, düğmeye tekrar 2 saniye basılı tutun ve bırakın. Mod/Durum Işığı turuncu yanıp sönerkenyeşil Zamanlama Işıklarının üçü yanıp söner, bu durum cihazın Ekstra Güç modunda olduğunu gösterir.
• Polimerizasyon için Güç Düğmesine basın. Zaman aralığı tamamlanmadan polimerizasyonu durdurmak için tekrar Güç Düğmesine basın.
• Standart Güç moduna dönmek için Süre/Mod Düğmesine 2 saniye basılı tutun ve bırakın. Mod/Durum Işığı yeşil yanarken yeşil Zamanlama Işıkları yanar, bu durum cihazın Standart
Güç modunda olduğunu gösterir.

Uyku Modu: Işıklı dolgu cihazı ile 60 saniye boyunca işlem yapılmadığında cihaz Uyku Moduna geçer, bu durum mod/durum ışığının yavaşça yanıp sönmesiyle belirtilir. Cihaz
kaldırıldığından veya cihaza temas edildiğinde güç tasarruf modundan çıkar ve otomatik olarak son kullanılan ayara döner. Pil ömrünü en üst düzeye çıkarmak için, kullanılmadığı zamanışıklı dolgu cihazını dokunmayın veya hareket ettirmeyin.

Temizleme

1) Kullanılmış bariyer kılıflarını her hastadan sonra standart atık olarak bertaraf edin.
2) İşleme Bölümüne bakınız.

Montaj Braketi Talimatları

1) Braket düz ve yağsız bir yüzeye monte edilmelidir.
2) Yüzeyi alkolle ovarak temizleyin.
3) Braketin yapışkan bantını geriye doğru soyun.
4) Braketi, ışıklı dolgu cihazı yukarı doğru çekilerek çıkarılacak şekilde yerleştirin. Sıkıca bastırarak yerine oturtun.

289

Hızlı Mod Kılavuzu:
Standart Güç
1000 mW/cm2

Mod

Yüksek Güç Plus
1600 mW/cm2

Ekstra Güç
3200 mW/cm2

Güç Düğmesi

Piller nasıl şarj edilir:1) Şarj cihazını ve pilleri sıvılar veya dezenfektan spreyler ile temas etmeyecekleri bir yere yerleştirin.
2) Pilleri, artı (+) ucu şarj cihazındaki gösterge ışıklarına bakacak şekilde şarj cihazına yerleştirin.
3) Yeşil Işıklar pillerin kullanıma hazır olduğunu gösterir.
4) Pillerin şarj olması 1-3 saat sürer. Pilleri kullanıma hazır olana kadar şarj cihazında bırakın.
NOT: Piller üç saatten uzun süre şarj edildiğinde şarj cihazındaki kırmızı ışık yeşile dönmezse, pil bozulmuş (arızalanmış) olabilir ve şarj edilemez. Yeni bir pil deneyin veya yeni bir şarjedilebilir pil seti sipariş etmek için Ultradent Müşteri Hizmetlerini arayın.

Mod/Zamanlama LED’leri

Süre Düğmeleri

Süre Seçenekleri

5s

10s

15s

20s

1s

2s

3s

Süreyi Değiştirmek için

Süre seçenekleri arasında geçiş yapmak için Süre Düğmesine hızlıca basıp bırakın.

Mod Değiştirmek için

Süre Düğmesine 2 saniye basılı tutun ve bırakın. Cihaz bir sonraki Moda geçecektir.

Semboller

Sabit LED'ler

4s

3s OnlyŞarj Süresi ve Pil Ömrü: Işık cihazında tam şarjlı pil ömrü, Mod/Zaman aralığına, pil tipine, kullanım miktarına ve LED verimliliğine bağlıdır.
• Önerilen Şarj Aralığı: Gerektiği gibi veya düşük pil göstergesi yandığında şarj edin. Yaşam beklentisi kullanıma bağlıdır.
• Ekstra Piller: Işık cihazı 4 adet şarj edilebilir pil ile birlikte gelir.

Kürleme Modu

Güç* (mW)Işınım* (mW/cm2)

Toplam Pozlama Süresi (Saniye)

Enerji** (Joule)Standart

970

900

5/10/15/20

4.8J/9.7J/14.5J/19.4J

Yüksek Güç Plus

1,620

1,500

1/2/3/4

1.6J/3.2J/4.8J/6.5J

Xtra

2,260

2,100

3

6.8J

*Lensin ucundan kompozitin üst yüzeyine 2 mm'lik bir mesafede nominal çıkış.
**Döngü başına toplam enerji (joule) bölümündeki değerler en yakın onda bir değere yuvarlanır.

Güç Seviyesi Uyarısı

Sıcaklık Uyarısı

Pilleri değiştirinSoğumaya bırakın• Düşük pil: yavaş yanıp sönme• Kapanma: 3 bip sesi, yanıp sönme• Çalışmaya izin verilmiyor• 3 bip sesi
• Yavaş yanıp sönme• Çalışmaya izin verilmiyor

Servis Uyarısı

LED Uyarısı

Onarım için Müşteri Hizmetlerini Arayın Onarım için Müşteri Hizmetlerini Arayın• Ses yok
• 2 saniyede bir yanıp sönme• Çalışmaya izin veriliyor

Piller nasıl değiştirilir/takılır:1) Saat yönünün tersine çeyrek tur çevirerek arka kapağı çıkarın.
2) Pilleri çıkarın.
3) Yeni pilleri önce artı (+) tarafa takın.
4) Saat yönünde çevirirken hizalayarak ve hafifçe iterek arka kapağı yeniden takın. Kapak tam olarak takıldığında tık sesi çıkaracaktır.
5) Ünite kullanıma hazırdır.
Düşük Piller: Düşük Pil Gösterge Işığı kırmızı renkte yanıp söndüğünde, ışık cihazı, kullanıcıya pilleri değiştirme zamanının geldiğini bildirir. Pil şarjı çok düşük olursa, sesli 3 bip uyarı sesi
duyulur ve ışık cihazı, piller yeniden şarj edilene veya yeni piller takılana kadar daha fazla çalışmaya izin vermez. (Bkz. Hızlı Uyarı Kılavuzu)

Yanıp sönen LED'ler

Hızlı Polimerizasyon Kılavuzu:

Hızlı Uyarı Kılavuzu::

2) Işık ölçerler büyük farklılıklar gösterir ve belirli ışık kılavuzu ipuçları ve lensler için tasarlanmıştır. Ultradent, Standart Güç modunda çıkışı rutin olarak kontrol etmenizi önerir. NOT:
gerçek sayısal çıktı, yaygın ışık ölçerlerin yanlışlığı ve ışık cihazındaki özel LED paketi nedeniyle çarpık olacaktır
Pil Bakımı
Pillerin Şarj Edilmesi ve Değiştirilmesi
Işık cihazı, 4 adet şarj edilebilir lityum demir fosfat pil ile birlikte gelir. Ultradent, yalnızca VALO şarj edilebilir pillerin kullanılmasını önerir.

• Sürekli 3 bip sesi
• Tek hızlı flaş• Çalışmaya izin verilmiyorÜretici Onarımı
1) Onarımlar sadece yetkili servis personeli tarafından yapılmalıdır. Ultradent, servis personeline onarımları gerçekleştirmeleri için belgeler sağlayacaktır.
2) Üniteleri onarım, servis veya kalibrasyon için gönderirken pilleri her zaman ışık cihazından ve şarj cihazından çıkarın. Pilleri, şarj cihazını, adaptörü ve ışık cihazını iade kutusuna ayrı
ayrı sarın.
3) Pilleri yerel yönetmeliklere uygun olarak gönderin.
Garanti
Ultradent Products, Inc. ("Ultradent"), bu ürünün satın alma tarihinden itibaren 5 yıl süreyle, ürünle birlikte verilen kullanım talimatlarına göre çalıştırıldığında, (i) Ultradent'in ürünle
birlikte verilen teknik belgelerinde belirtilen özelliklere her açıdan uygun olacağını ve (ii) malzeme ve işçilik kusurları içermeyeceğini garanti eder.
Bu sınırlı garanti devredilemez ve yalnızca orijinal alıcı için geçerlidir ve ürünün sonraki sahiplerini kapsamaz. Bu sınırlı garanti piller, şarj cihazları, adaptörler veya uyarlanabilir lensler
gibi ancak bunlarla sınırlı olmamak üzere diğer aksesuar bileşenlerini kapsamaz. Bu sınırlı garanti, ürünün ihmal, kötüye kullanım, yanlış kullanım, kaza, modifikasyon, kurcalama,değişiklik veya geçerli kullanım talimatlarına uyulmaması nedeniyle arızalanması veya hasar görmesi durumunda geçersizdir. Yalnızca örnek olması amacıyla, düşürülen ve hasar gören
bir ürün bu garanti kapsamında değildir. Bu sınırlı garanti kapsamında hak kazanmak için, satın alma kanıtı (örneğin, satış fişi veya benzer belgeler) kusurlu ürünle birlikte Ultradent'e
sunulmalıdır.
Burada belirtilen garanti koşullarını karşılayan kusurlu bir ürün, tamamen Ultradent'in takdirine bağlı olarak onarılacak veya değiştirilecektir. Ultradent'in ürünle ilgili sorumluluğu hiçbir
durumda alıcı tarafından ödenen satın alma fiyatını aşamaz. Ultradent hiçbir koşulda, bu ürünün kullanımından kaynaklanan veya bununla bağlantılı olarak ortaya çıkan dolaylı, arızi,öngörülen, öngörülemeyen özel veya sonuç olarak ortaya çıkan zararlardan sorumlu tutulamaz.

İşlemden Geçirme

Bu ürün bir bariyer kılıfı ile kullanılmak üzere tasarlanmıştır. Bakım ve genel temizlik için yukarıya bakın.

Saklama ve BertarafIşıklı dolgu cihazı 2 haftadan uzun bir süre boyunca saklanacaksa ya da taşımak üzere paketlenecekse mutlaka pilleri çıkarın. Piller şarj edilmeden uzun bir süre boyunca cihazın içinde
bırakılırsa çalışmaz veya şarj edilemez hale gelebilirler. Pilleri 40°C (104 °F) üzeri sıcaklıklarda veya doğrudan güneş ışığı alan yerlerde muhafaza etmeyin.
Işıklı Dolgu Cihazının Saklanması ve Taşınması:• Sıcaklık: +10 °C – +40 °C (+50 °F – +104 °F)• Bağıl Nem: %10 ila %95• Ortam Basıncı: 500 hPa ila 1060 hPa

BakımIşık Cihazının Genel Temizliği
Her kullanımdan sonra, bir gazlı bezi veya yumuşak bir bezi onaylanmış bir yüzey dezenfektanı ile nemlendirin ve yüzeyi ve lensi silin. Onaylanmamış temizleyiciler ışık cihazına zarar
verebilir.
KABUL EDİLEBİLİR TEMİZLEYİCİLER:• %70 İzopropil alkol• %70 EtanolIşık Kalkanının Genel Temizliği:VALO Cordless ışık kalkanını herhangi bir yüzey dezenfektanı kullanarak temizleyin. Otoklavlama YAPMAYIN.
OnarımKullanıcı Tarafından Gerçekleştirilen Bakım
1) Dental kompozitin lensin yüzeyine yapışmasını önlemek için bir bariyer kılıfı kullanın. Gerekirse, yapışmış kompozitleri dikkatlice çıkarmak için plastik veya paslanmaz çelik bir diş aleti
kullanın. Lense zarar verecek aletler kullanmayın.
290

Elektronik atıkları (ör. ışıklı dolgu cihazları, şarj cihazları, piller ve güç kaynakları) bertaraf ederken yerel atık ve geri dönüşüm yönergelerine uyun.

Teknik Faktörler
AksesuarlarÜrün

CE Uygunluk Bilgisi

VALO Kablosuz Işık Kalkanı

291

Ürün

CE Uygunluk Bilgisi

Teknik Bilgiler/Veriler

Etkili Kompozit Polimerizasyon Dalga Boyu Bantları:

VALO Şarj Cihazı

ETKILI KOMPOZIT KÜRLEME DALGA BOYU BANTLARI(Kullanılabilir Dalga Boyu Aralığı: 385-515 nm)

SPEKTRAL RADYANT GÜÇ (mW/nm)

VALO Piller

Etkili düşük
dalga boyu bandı
395-415 deniz mili

Etkili üst dalga
boyu bandı
440-480 deniz mili

Xtra
Yüksek GüçStandart

DALGAENGİNLİĞİ (nm)Nitelik

Bilgi/Teknik ÖzellikLensÇap 11,7 mm

Dalga boyu aralığı

• Kullanılabilir dalga boyu aralığı: 385 – 515nm• Tepe dalga boyları: 395 – 415nm ve 440 – 480nm
Nominal Işınım Uyarıcılık Karşılaştırma Tablosu
Ölçüm AletiIşık Şiddeti Tablosu

292‡ Gigahertz spektrum analizörü
Uyarıcılık

Toplam Güç

Metre Açıklığı

15 mm

15 mm

Standart Güç(±%10)

900 mW/cm²

970 mW

Yüksek Güç Plus(±%10)

1500 mW/cm²

1615 mW

Ekstra Güç (±%10)2100 mW/cm²

Radyant Çıkış, cihaz kapasitesine, ölçüm
yöntemine ve ışık yerleşimine bağlı olarakdeğişecektir.
‡ Radyant Çıkış, Gigahertz-Optik spektrum
analizörü ile ölçüldüğünde ISO 10650'yeuygundur.

2260 mW

VALO Grand Işıklı
Dolgu Cihazı

Uyumluluk: IEC 60601-1 (Güvenlik), IEC 60601-1-2(Elektromanyetik Uyumluluk-EMC)Ağırlık:• Pillerle: 6 ons (190 gram)• Piller olmadan: 5 ons (150 gram)
Boyut: (203 x 32,5 x 27) mm, (8 x 1,28 x 1,06) inçŞarj Cihazı Güç
Kaynağı

Çıkış – 500mA'da 12VDC
Giriş – 100VAC ila 240VAC
Evrensel fiş adaptörlerine sahip Ultradent P/N 5930
VALO Şarj Cihazı Güç Kaynağı

Uyumluluk: IEC 60601-1 (Güvenlik)
Kablo uzunluğu – 1,8 metre/6 fit
VALO Şarj Cihazı Güç Kaynağı, Tıbbi Standartlarda Sınıf II bir güç kaynağıdır ve
ANA güç kaynağından izolasyon sağlar

VALO Şarj Cihazı

VALO 3,6 VDC Lityum Demir Fosfat akıllı pil şarj cihazı:• Tamamen şarj olduğunda otomatik kapanma• Bozuk pillerin otomatik olarak tespit edilmesi• Korumalar: Termal, Aşırı şarj, Kısa devre, ters polarite
⸰ Kırmızı LED - Şarj Oluyor⸰ Yeşil LED – Boş veya Tamamen Şarj Oldu⸰ LED kapalı – kısa devre• Şarj süresi: 1 ila 3 saat
Uyumluluk: CE, WEEE
293

Nitelik

Bilgi/Teknik Özellik

Elektromanyetik Emisyonlara İlişkin Kılavuz ve Üreticinin Beyanı

VALO PillerŞarj edilebilir: Kimyasal olarak güvenli Lityum Demir Fosfat (LiFePO4) RCR123A• Çalışma Gerilimi: 3,2 VDC
Uyumluluk: UL, CE, RoHS, WEEE

RF emisyonları
CISPR 11

Grup 1Işıklı dolgu cihazı pille çalışır ve EMI, RF veya aşırı gerilimden etkilenmez.

RF emisyonu
CISPR 11B Sınıfı

Çalışma Koşulları

Sıcaklık: +10 °C ila +32 °C (+50 °F ila +90 °F)Bağıl Nem: %10 ila %95
Ortam Basıncı: 700 hPa ila 1060 hPaIşıklı dolgu cihazı elektriksel ve elektromanyetik enerjiyi sadece kendi işlevleri için kullanır. Bu
nedenle, cihazın RF emisyonu çok düşüktür ve yakınlardaki elektronik ekipmanlarda parazite neden
olmayacaktır.

Geçerli Değil

Işıklı Dolgu Cihazı kısa süreli kullanım için tasarlanmıştır. Maksimum ortam sıcaklığında (32 °C) 1 dakika AÇIK arka arkaya döngü, 30
dakika KAPALI (soğuma süresi).

Harmonik emisyonlar
IEC 61000-3-2

Harmonik emisyon ve gerilim dalgalanması testi, ışıklı dolgu cihazına uygulanmaz çünkü cihaz
pille çalışır.

Çalışma Döngüsü

Gerilim dalgalanmaları/titreşim
yayılımları IEC 61000-3-3

Geçerli Değil

Işıklı dolgu cihazı, konutlar ve evsel amaçla kullanılan binalara elektrik tedarik eden kamuya
ait düşük gerilimli elektrik şebekesine doğrudan bağlı olan tesisler de dâhil olmak üzere tüm
tesislerde kullanıma uygundur.

Sorun GidermeAşağıda önerilen çözümler sorunu gidermezse lütfen 800.552.5512 numaralı telefondan Ultradent'i arayın. ABD dışında iseniz Ultradent distribütörünüzü veya dentalürün satıcınızı arayın.
Sorun

Olası Çözümler
1) Güç Tasarruf Modundan çıkarmak için Süre/Mod Düğmesine veya Güç Düğmesine basın.
2) Pil şarj durumu için kırmızı Düşük Pil Seviyesi Göstergesini kontrol edin.
3) Dolu pillerin cihaza doğru şekilde takılıp takılmadığını kontrol edin.
4) Kırmızı ve sarı Uyarı LED’leri yanıp sönüyorsa bu ışıklı dolgu cihazının iç sıcaklık güvenlik sınırına ulaştığı anlamına gelir.
Cihazı 10 dakika kadar soğumaya bırakın veya üniteyi hızlı bir şekilde soğutmak için serin nemli bir havlu kullanın.
5) Kırmızı Uyarı LED’i sürekli olarak yanıp sönüyor ve bip sesi çıkarıyorsa onarım için Ultradent Müşteri Hizmetleri’ni arayın.

Işık yanmıyorIşık istenilen süre boyunca yanık
kalmıyor

1) Mod ve Zamanlama ışıkları için süre girişinin doğru olup olmadığını kontrol edin.
2) Pil şarj durumu için Düşük Pil Seviyesi Göstergesini kontrol edin.
3) Dolu pillerin cihaza düzgün takılıp takılmadığını kontrol edin.

Işık rezini düzgün polimerize etmiyor

1) Lenste polimerize rezin/kompozit artığı kalıp kalmadığını kontrol edin.
2) Bu işleme uygun, kehribar UV korumalı gözlük kullanarak LED ışıklarının çalışıp çalışmadığını kontrol edin.
3) Işıkölçerle güç seviyesini kontrol edin. Işıkölçer kullanacaksanız Ultradent, ışıklı dolgu cihazını Standart Güç modunda
kontrol etmenizi önermektedir.
NOTE: The true numeric output will be skewed due to the inaccuracy of common light meters and the custom LED pack the
curing light uses. Light meters differ greatly and are designed for specific light guide tips and lenses.
4) Polimerize rezinin son kullanma tarihini kontrol edin.
5) Üreticinin talimatlarına uygun tekniğin kullanıldığından emin olun.

Piller şarj olmuyor

1) Pillerin şarj cihazına doğru yönde takıldığından emin olun ve pillerin şarj olması için 1-3 saat bekleyin.
2) Şarj cihazındaki kırmızı ışıklar yeşile dönmezse, yeni pil ve/veya şarj cihazı sipariş etmek için Ultradent Müşteri
Hizmetlerini arayın.
3) Şarj cihazında yeşil ışık da kırmızı ışık da yanmıyorsa, yeni bir şarj cihazı ve/veya AC adaptör sipariş etmek veya
elinizdekileri değiştirmek için Ultradent Müşteri Hizmetlerini arayın.

Şarj cihazı pilleri şarj etmiyor1) Şarj cihazının takılı ve AC adaptörün çalışan bir elektrik prizine takılı olduğundan emin olun.
2) Şarj cihazında yeşil ya da kırmızı ışıklar yanmıyorsa, yeni bir şarj cihazı ve/veya AC adaptör için Ultradent Müşteri
Hizmetleri’ni arayın.

Mod veya zaman aralıklarıdeğiştirilemiyor1) Işıklı dolgu cihazının kilidinin açıldığını gösteren bir dizi bip sesi duyana kadar hem Süre/Mod hem de Güç düğmelerine
basılı tutun.

Çeşitli BilgilerIşıklı dolgu cihazı aşağıda belirtilen elektromanyetik ortamda kullanılmak üzere tasarlanmıştır. Müşteri ya da kullanıcı, cihazın böyle bir ortamda kullanıldığından emin
olmalıdır.

294

Işıklı dolgu cihazı aşağıda belirtilen elektromanyetik ortamda kullanılmak üzere tasarlanmıştır. Müşteri ya da kullanıcı, cihazın böyle bir ortamda kullanıldığından emin
olmalıdır.
BAĞIŞIKLIK testi

IEC 60601 test düzeyi

Uygunluk düzeyi

Elektromanyetik ortam kılavuzu

Elektrostatik boşalma (ESD)± 8 kV temas± 15 kV hava

± 8 kV temas± 15 kV hava

Fiziksel çevre aşağıdakilerle sınırlandırılmalıdır:1) IP Kodu: IP20
2) Sıvıya batırmayın.
3) Yanıcı gaz etrafında kullanmayın. Cihaz, APG ve AP
kategorisinde değildir.
4) Depolama nem aralığı: %10 – %95
5) Depolama sıcaklık aralığı: 10 °C – 40 °C

Elektriksel hızlı geçici rejim/
patlamaIEC 61000-4-4

Güç hatları için ± 2 kVGiriş/çıkış hatları için ± 1 kV

Güç hatları için ± 2 kV
Not 1: Işıklı dolgu cihazında I/0
portu bulunmamaktadırIşıklı dolgu cihazı pille çalışır ve AC ŞEBEKE elektriğine
bağlanamaz.

Ani gerilim yükselmesi
IEC 61000-4-5

Faz-faz ± 1 kV
Faz-toprak ± 2 kV

Faz-faz ± 1 kV
Faz-toprak ± 2 kV

Güç kaynağı giriş hatlarında
gerilim düşmeleri, kısa kesintilerve voltaj değişimleri<%5 U(0,5 döngüde U'da >%95 düşüş)<%5 U(0,5 döngüde U'da >%95 düşüş)%40 U(5 döngüde U'da %60 düşüş)%40 U(5 döngüde U'da %60 düşüş)%70 U(25 döngüde U'da %30 düşüş)%70 U(25 döngüde U'da %30 düşüş)<%5 U(5 saniyede U'da >%95 düşüş)<%5 U(5 saniyede U'da >%95 düşüş)Not 2: Kendi kendine düzelir

30 A/m30 A/mIEC 61000-4-2

IEC 61000-4-11

Güç frekansı(50/60 Hz) manyetik alan
IEC 61000-4-8

Elektromanyetik Emisyonlara İlişkin Kılavuz ve Üreticinin Beyanı

Emisyon Testi

Elektromanyetik Bağışıklığa İlişkin Kılavuz ve Üreticinin Beyanı

UygunlukIşıklı dolgu cihazı pille çalıştığı için, AC ŞEBEKE
elektriğinde meydana gelen elektrik geçişleri, aşırı
gerilim, gerilim düşmeleri, kısa kesintiler veya voltajdeğişimlerine maruz kalmaz.
Aksesuar şarz cihazı yukarıdaki durumlara maruz
kalabilir ancak cihazdan bağımsızdır ve VALO Kablosuzışıklı dolgu cihazının çalışmasını etkilemez.
Işıklı dolgu cihazının pil voltajı 4 VDC'ye düşerse cihazçalışmayacaktır. VALO Kablosuz ışıklı dolgu cihazı
kapanacaktır. Yeni piller takıldığında ve uygun güç
seviyeleri eski durumuna döndüğünde, ışıklı dolgu
cihazı yeniden başlayacak ve elektrik kesintisindenönceki durumuna dönecektir. Güç kaybı durumundaışıklı dolgu cihazı kendiliğinden düzelir.
Güç frekansı manyetik alanları tipik bir konut, evdesağlık ortamı, ticari bina veya hastanede ya da askeri
ortamda bulunan tipik bir yere özgü seviyelerde
olmalıdır.

Not 1: Işıklı dolgu cihazında herhangi bir port veya herhangi bir erişilebilir G/Ç hattı bulunmamaktadır.

Elektromanyetik ortam - kılavuz

295

Hayat dışı destek sistemleri için Elektromanyetik Bağışıklığa İlişkin Kılavuz ve Üreticinin BeyanıTaşınabilir ve mobil RF iletişim aracı ile VALO Kablosuz ışıklı dolgu cihazı arasındaki önerilen ayırma mesafeleri için Kılavuz ve Üreticinin BeyanıIşıklı dolgu cihazı aşağıda belirtilen elektromanyetik ortamda kullanılmak üzere tasarlanmıştır. Müşteri ya da kullanıcı, cihazın böyle bir ortamda kullanıldığından emin
olmalıdır.

Işıklı dolgu cihazı yayılan RF bozulmalarının kontrol edildiği bir elektromanyetik ortamda kullanılmak üzere tasarlanmıştır. Işıklı dolgu cihazının kullanıcısı, iletişim
aracının maksimum çıkış gücüne göre, taşınabilir ve mobil RF iletişim aracı (vericiler) ve cihaz arasında aşağıda tavsiye edilen minimum mesafeyi koruyarak
elektromanyetik parazitin önlenmesine yardımcı olabilir.

BAĞIŞIKLIK testi

IEC 60601 test düzeyi

Uygunluk düzeyi

İletilen RF

3 Vrms3 VrmsIEC 61000-4-6

150 kHz ila 80 MHz

150 kHz ila 80 MHz

Elektromanyetik ortam kılavuzuTaşınabilir ve mobil RF iletişim araçları, kablolar da dâhil olmak üzere ışıklı dolgu cihazının
hiçbir parçasına, verici frekansı için geçerli denklemle hesaplanan önerilen ayırma
mesafesinden daha yakında kullanılmamalıdır.
Önerilen ayırma mesafesi:

80 MHz ila 800 MHz

Yayılan RF

3 V/m

3 V/m

800 MHz ila 2,5 GHzP, vericinin üreticisinin belirlediği Watt (W) cinsinden maksimum çıkış gücü derecesi, d ise
metre (m) cinsinden önerilen ayırma mesafesidir.
Sabit RF vericilerin elektromanyetik alan araştırmasıa ile belirlenen alan şiddeti, her frekansaralığında uygunluk seviyesinden düşük olmalıdırb.

IEC 61000-4-3

80 MHz ila 2,5 GHz

80 MHz ila 2,5 GHzÜzerinde şu sembolün bulunduğu cihazların yakınında kullanıldığında parazite neden
olabilir:

Vericinin frekansına göre ayırma mesafesi(metre)Vericinin
maksimum nominal çıkış gücü
(Watt cinsinden P)

150 kHz – 80 MHz

80 MHz – 800 MHz

800 MHz – 2.5 GHz

0,01

0,12 metre

0,035 metre

0,07 metre

0,1

0,37 metre

0,11 metre

0,22 metre

1

1,7 metre

0,35 metre

0,7 metre

10

3,7 metre

1,11 metre

2,22 metre100

11,7 metre

3,5 metre

7,0 metreIşıklı dolgu cihazı IEC 60601-1-2:2014'e göre test edilmiş ve 80 kHz ila 2,5 GHz arasında 10V/m yayılan alan şiddeti altında testi geçmiştir. Yukarıdaki formüllerde 3
Vrms değeri V1'e, 10V/m değeri ise E1'e karşılık gelir.
Yukarıdaki listede yer almayan maksimum çıkış gücüne sahip vericiler için; metre cinsinden (m) önerilen ayırma mesafesi, vericinin frekansı için geçerli denklem
kullanılarak hesaplanabilir. Burada P, vericinin üreticisinin belirlediği watt (W) cinsinden maksimum çıkış gücü derecesidir.
NOT 1: 80 MHz ve 800 MHz'de, daha yüksek frekans aralığı için ayırma mesafesi geçerlidir.
NOT 2: Bu yönergeler her durumda geçerli olmayabilir. Elektromanyetik dalga yayılımı; yapılar, nesneler ve insanlara bağlı olarak oluşan soğurulma ve yansımadan
etkilenir.

NOT 1: 80 MHz ve 800 MHz’de daha yüksek frekans aralıkları geçerlidir.
NOT 2: Bu yönergeler her durumda geçerli olmayabilir. Elektromanyetik dalga yayılımı; yapılar, nesneler ve insanlara bağlı olarak oluşan soğurulma ve yansımadan
etkilenir.
aRadyo (hücresel/kablosuz) telefonlar ve kara mobil radyoları, amatör radyolar, AM ve FM radyo yayını ve TV yayınına yönelik baz istasyonları gibi sabit vericilerinalan şiddeti teorik olarak doğru bir şekilde tahmin edilemez. Sabit RF vericilerinden kaynaklanan elektromanyetik ortamı değerlendirmek için bir elektromanyetik
alan araştırması düşünülmelidir. VALO Kablosuz ışıklı dolgu cihazının kullanıldığı konumda ölçülen alan şiddeti yukarıda belirtilen geçerli RF uygunluk düzeyini aşarsa,
cihazın normal çalışıp çalışmadığı gözlemlenmelidir. Anormal bir performans gözlemlenirse, ışıklı dolgu cihazının yönünü veya yerini değiştirme gibi ek önlemler
gerekebilir.
b
150 kHz – 80 MHz frekans aralığında, manyetik alan şiddeti 3 V/m’den düşük olmalıdır.

296

297

‫‪AR‬‬‫وصف المنتج‬

‫	••‬
‫	••‬
‫	••‬

‫™‪ VALO‬جراند‬‫تعليمات االستخدام‬‫بفضل طيفه عريض النطاق‪ ،‬تم تصميم ضوء المعالجة ‪ VALO‬جراند لبلمرة جميع المنتجات المعالجة بالضوء في نطاق الطول الموجي من ‪ 515-385‬نانومتر وف ًًقا لمعيار ‪ .ISO 10650‬يستخدم ضوء المعالجة ‪VALO‬‬‫جراند بطاريات قابلة إلعادة الشحن وشاحن بطارية من ‪ .Ultradent VALO‬تم تصميم ضوء المعالجة ليوضع في حامل وحدة طب األسنان القياسي أو يمكن تركيبه بشكل مستقل باستخدام الحامل المضمن في‬‫العلبة‪.‬‬‫مكونات المنتج فالو جراند‪:‬‬‫	•‪ – 1‬ضوء المعالجة ‪ VALO‬جراند‬‫	•‪ – 4‬بطاريات ‪ Ultradent VALO‬القابلة إلعادة الشحن‬‫	•‪ – 1‬شاحن بطارية ‪ Ultradent VALO‬مزود بشاحن طاقة طبي ‪ 12‬فولت تيار مستمر وتيار متناوب‬‫	•‪ - 1‬مجموعة نماذج غالف التكميم ‪VALO‬‬‫	•‪ – 1‬الواقية الضوئية ل ‪ VALO‬جراند‬‫	•‪ – 1‬دعامة ضوء المعالجة للتثبيت عىل السطح بشريط الصق مزدوج‬‫نظرة عامة عىل مفاتيح التحكم‪:‬‬‫مفاتيح الرموز‬‫(‪1‬اشحن البطاريات قبل استخدام جهاز التصليب الضوئي (انظر قسم صيانة البطارية)‪.‬‬‫(‪2‬ثبِّت جهاز التصليب الضوئي في سناد تثبيت وحدة األسنان القياسية أو سناد تثبيت الملحقات حتى يكون جاهزًا لالستخدام‪.‬‬‫(‪3‬قبل كل استخدام‪ ،‬قم بارتداء قفازات طبية‪.‬‬‫ارتداء القفازات الطبية‪:‬‬‫مالحظة‪:‬‬‫	•يؤدي استخدام القفازات الطبية إىل تقليل ناتج الضوء بنسبة ‪ .10٪-5‬نظرًا الرتفاع خرج الطاقة لجهاز التصليب الضوئي‪ ،‬فقد ثبت أن التصليب متكافئ بدرجة كبيرة‪.‬‬‫	•يجب تنظيف جهاز التصليب الضوئي وتطهيره باستخدام مواد تنظيف و‪ /‬أو تطهير مناسبة بعد استخدامه مع كل مريض‪ .‬انظر القسم الوارد تحت عنوان "المعالجة"‪.‬‬‫واقي مصباح فالو كوردليس‪:‬‬‫زر (أزرار) التشغيل‬‫(الجزء العلوي أو السفلي)‬‫إن واقي مصباح فالو كوردليس عىل شكل بيضاوي‪ ،‬ويمكن تدويره لتحقيق أقصى استخدام‪ ،‬ويمكن استخدامه مع غطاء حاجز شفاف‪.‬‬‫االستخدام‬‫البطارية منخفضة ‪ -‬مؤشر‬‫(‪1‬يتم استخدام كل وضع تشغيل لتصليب مواد األسنان باستخدام البادئات الضوئية‪ .‬انظر دليل الوضع السريع لمعرفة أوقات التصليب الموصى بها‪.‬‬‫مالحظة‪ :‬جهاز التصليب الضوئي ُمبرمج لالنتقال من وضع التشغيل القياسي إىل وضع التشغيل المرتفع باإلضافة إىل وضع التشغيل الفائق عىل التوالي‪ .‬عىل سبيل المثال‪ ،‬لالنتقال من وضع التشغيل القياسي‬‫إىل وضع التشغيل الفائق‪ ،‬من الضروري االنتقال إىل وضع التشغيل المرتفع ثم إىل وضع التشغيل الفائق‪.‬‬‫استخداما مؤخرًا‪ ،‬وسيعود إىل هذا الوضع افتراض ًيا عند تغيير األوضاع أو في حالة نزع البطاريات‪.‬‬‫(‪2‬يقوم جهاز التصليب الضوئي بتسجيل الفاصل الزمني والوضع األكثر‬‫ً‬‫زر تغيير التوقيت‪ /‬الوضع‬‫التشغيل‬‫وضع التصليب‪ :‬وضع التشغيل القياسي‬‫مصابيح التوقيت‬‫مصباح الوضع‪ /‬الحالة‬‫(الجزء العلوي)‬‫(الجزء السفلي)‬‫بالنسبة لجميع المنتجات المذكورة‪ ،‬يُرجى قراءة وفهم جميع التعليمات والمعلومات الواردة في بطاقات بيانات السالمة بعناية قبل االستخدام‪.‬‬‫دواعي االستخدام‪ /‬األغراض المنشودة‬‫الحساسة للضوء والمواد الالصقة‪.‬‬‫مصدر اإلضاءة لتصليب مواد ترميم األسنان‬‫ّ‬‫التحذيرات واالحتياطات‬‫مجموعة المخاطر ‪2‬‬‫تنبيه من األشعة فوق البنفسجية المنبعثة من هذا المنتج؛ فقد ينجم عن التعرض لها تهيج العين أو الجلد‪ .‬استخدِم أدوات الحماية المناسبة‪.‬‬‫تنبيه من اإلشعاع البصري الخطير عىل نحو محتمل المنبعث من هذا المنتج‪ ،‬فتجنب التحديق في مصباح التشغيل؛ فقد يُلحق ذلك ضررًا بالعين‪.‬‬‫دائما ارتداء نظارات واقية للعين من األشعة فوق البنفسجية بلون الكهرمان عند استخدام ضوء المعالجة‪.‬‬‫	•ال تنظر مباشرة إىل مخرج انبعاث الضوء‪ .‬يجب عىل المريض والطبيب والمساعدين ً‬‫	•ال يُسمح بتعديل هذا الجهاز لمنع خطر حدوث صدمة كهربائية‪ .‬استخدم فقط مصدر الطاقة ‪ Ultradent VALO‬المرفق ومحوالت القابس‪ .‬في حالة تلف هذه المكونات‪ ،‬ال تستخدم الجهاز واتصل بخدمة عمالء‬‫‪ Ultradent‬لطلب بديل‪.‬‬‫	•قد تؤدي معدات االتصاالت الالسلكية المحمولة إىل خفض مستوى األداء إذا تم استخدامها عىل مسافة أقرب من ‪ 30‬سم (‪ 12‬بوصة)‪.‬‬‫	•استخدم فقط الملحقات والكابالت وإمدادات الطاقة المصرح بها لمنع التشغيل غير السليم أو زيادة االنبعاثات الكهرومغناطيسية أو انخفاض المناعة الكهرومغناطيسية (راجع قسم االنبعاثات الكهرومغناطيسية)‪.‬‬‫	•لتجنب خطر نشوب حريق كهربائي مرتبط بالتعامل مع البطاريات‪:‬‬‫	⸰ال تخلط بين البطاريات القابلة إلعادة الشحن مع البطاريات غير القابلة إلعادة الشحن أو أنواع البطاريات األخرى‪.‬‬‫	⸰ال تحاول شحن البطاريات غير القابلة إلعادة الشحن‪ .‬يمكن أن يؤدي شحن البطاريات غير القابلة إلعادة الشحن إىل إصابة أو تلف الممتلكات‪.‬‬‫	⸰ال تقم بتعقيم أو رش البطاريات أو أقطاب البطارية أو الشاحن أو محول طاقة التيار المتردد بأي نوع من أنواع السوائل‪ .‬إذا ظهر التآكل عىل أقطاب شاحن البطارية‪ ،‬فاتصل بخدمة عمالء ‪ Ultradent‬لطلب‬‫بديل‪.‬‬‫	⸰ال تشحن البطاريات في مناطق قريبة من المواد القابلة لالشتعال‪.‬‬‫	⸰ال تبقي الشاحن في غرفة العمليات السريرية‪.‬‬‫	•لتجنب خطر اإلصابة‪ ،‬ال تستخدم البطاريات المتآكلة (الصدئة)‪ ،‬أو المنبعجة‪ ،‬أو التي تنبعث منها رائحة أو سوائل‪ ،‬أو التي تحتوي عىل غالف ممزق أو مفقود‪ ،‬أو التالفة بطريقة أخرى‪ .‬اتصل بخدمة عمالء‬‫‪ Ultradent‬لطلب بطاريات بديلة‪.‬‬‫	•استخدم البطاريات الموصى بها فقط‪ .‬يمكن أن تتسبب البطاريات البديلة في حدوث عطل‪.‬‬‫ثوان في أي وضع‪ .‬إذا كانت هناك حاجة إىل أوقات عالج أطول‪ ،‬فاستخدم دورات‬‫	•لمنع خطر التهيج الحراري أو اإلصابة‪ ،‬تجنب دورات المعالجة المتتالية وال تعرض األنسجة الرخوة الفموية عىل مقربة ألكثر من ‪10‬‬‫ٍ‬‫عالج متعددة مع فترات راحة بين الدورات أو استخدم منتجًا مزدوج المعالجة لتجنب تسخين األنسجة الرخوة‪.‬‬‫	•توخ الحذر عند عالج المرضى الذين يعانون من تفاعالت أو حساسيات ضوئية ضارة‪ ،‬أو المرضى الذين يخضعون للعالج الكيميائي‪ ،‬أو المرضى الذين يعالجون بأدوية حساسة للضوء‪.‬‬‫	•قد تكون هذه الوحدة عرضة لمجاالت كهربائية مغناطيسية أو ثابتة قوية‪ ،‬مما قد يعطل البرمجة‪ .‬إذا كنت تشك في حدوث ذلك‪ ،‬فافصل الوحدة مؤقتًا ثم أعد توصيلها بالمأخذ‪.‬‬‫	•ال تمسح ضوء المعالجة بالمنظفات الكاوية أو الكاشطة أو األوتوكالف أو تغمره في أي نوع من أنواع الغسول بالموجات فوق الصوتية أو المطهر أو محلول التنظيف أو السوائل‪ .‬قد يؤدي عدم اتباع تعليمات‬‫المعالجة المضمنة إىل تعطيل ضوء المعالجة‪.‬‬‫	•لتجنب إتالف الجهاز‪ ،‬ال تقم بإدخال أصابع أو أدوات أو أشياء أخرى في حجرة البطارية في ضوء المعالجة‪.‬‬‫	•لتجنب إتالف الجهاز‪ ،‬ال تحاول تنظيف نقاط التالمس الكهربائية‪ ،‬أو أي جزء من حجرة البطارية‪ .‬اتصل بخدمة عمالء ‪ Ultradent‬إذا كانت هناك مخاوف‪.‬‬‫‪298‬‬‫تعليمات تدريجية‬‫اإلعداد‬‫تم تصميم القفازات الطبية لتتالءم مع جهاز التصليب الضوئي وتحافظ عىل نظافة سطحه‪ .‬تساعد القفازات الطبية في الحيلولة دون حدوث التلوث الخلطي‪ ،‬وتساعد في منع المواد المركبة في طب األسنان من‬‫االلتصاق بسطح العدسة وجهاز التصليب‪ ،‬وتمنع كذلك تغير اللون والتآكل الناجمين عن محاليل التنظيف‪.‬‬‫صورة فوتوغرافية‬‫األزرار والمصابيح‬‫لمنع خطر التلوث المتبادل‪ ،‬تستخدم الحواجز المكممة لمريض واحد‪.‬‬‫لتقليل خطر التآكل‪ ،‬قم بإزالة الكم الحاجز بعد االستخدام‪.‬‬‫لتقليل خطر الراتنجات غير المعالجة‪ ،‬ال تستخدم ضوء المعالجة في حالة تلف العدسة‪.‬‬‫	•لتقليل خطر التآكل‪ ،‬يرجى نزع القفازات الطبية بعد االستخدام‪.‬‬‫	•لتقليل مخاطر الراتنجات المستخدمة في التصليب‪ ،‬تجنب استخدام جهاز التصليب الضوئي في حال تلف العدسة‪.‬‬‫الفترات الزمنية الفاصلة‪ 20 ،15 ،10 ،5 :‬ثانية‪.‬‬‫	•ينتقل جهاز التصليب الضوئي افتراض ًيا إىل هذا الوضع عندما يكون قيد التشغيل في البداية‪ .‬يُصدر مصباح الوضع‪ /‬الحالة ضوءًا أخضر وتضيء مصابيح التوقيت األربعة الخضراء‪ ،‬مما يشير إىل وضع‬‫التشغيل القياسي‪.‬‬‫	•لتغيير الفترات الزمنية‪ ،‬اضغط بسرعة عىل زر الوقت‪ /‬الوضع‪.‬‬‫	•اضغط عىل زر التشغيل للتصليب‪ .‬إليقاف التصليب قبل إكمال فترة زمنية فاصلة‪ ،‬اضغط عىل زر التشغيل مر ًة أخرى‪.‬‬‫وضع التصليب‪ :‬وضع التشغيل المرتفع الفائق‬‫ثوان‪.‬‬‫الفترات الزمنية الفاصلة‪4 ،3 ،2 ،1 :‬‬‫ٍ‬‫	•من وضع التشغيل القياسي‪ ،‬اضغط مع االستمرار عىل زر تغيير الوقت‪ /‬الوضع لمدة ثانيتين ثم حرره‪ .‬يُصدر مصباح الوضع‪ /‬الحالة ضوءًا برتقال ًيا‪ ،‬وتضيء مصابيح التوقيت األربعة الخضراء وتومض‪ ،‬مما‬‫يشير إىل وضع التشغيل المرتفع‪.‬‬‫	•لتغيير الفترات الزمنية‪ ،‬اضغط بسرعة عىل زر الوقت‪ /‬الوضع‪.‬‬‫	•اضغط عىل أي زر طاقة للتصليب‪ .‬إليقاف التصليب قبل إكمال فاصل زمني‪ ،‬اضغط عىل أي زر طاقة مرة أخرى‪.‬‬‫	•للعودة إىل وضع التشغيل القياسي‪ ،‬اضغط مع االستمرار عىل زر تغيير الوقت‪ /‬الوضع لمدة ثانيتين ثم حرره‪ ،‬وبذلك سيتم االنتقال إىل وضع التشغيل الفائق‪ .‬اضغط مع االستمرار مرة أخرى لمدة ثانيتين‪ ،‬ثم‬‫حرره‪ .‬يُصدر مصباح الوضع ‪ /‬الحالة ضوءًا أخضر وتضيء مصابيح التوقيت األربعة الخضراء‪ ،‬مما يشير إىل الوضع القياسي‪.‬‬‫وضع التصليب‪ :‬وضع التشغيل الفائق‬‫ثوان فقط (مالحظة‪ :‬يتأخر وضع التشغيل الفائق ثانيتين بغرض السالمة في نهاية كل دورة تصليب للحد من التدفئة أثناء التصليب المتتابع‪ .‬في نهاية فترة التأخير‪ ،‬يشير التنبيه الصوتي‬‫الفترة الزمنية الفاصلة‪3 :‬‬‫ٍ‬‫إىل أن الوحدة جاهزة لالستخدام المستمر‪.‬‬‫ً‬‫ووميضا‪ ،‬وتضيء‬‫	•من وضع التشغيل القياسي‪ ،‬اضغط عىل زر تغيير الوقت‪ /‬الوضع لمدة ثانيتين ثم حرره واضغط مع االستمرار مرة أخرى لمدة ثانيتين ثم حرره‪ .‬يُصدر مصباح الوضع‪ /‬الحالة ضوءًا برتقال ًيا‬‫ثالثة من مصابيح التوقيت الخضراء وتومض‪ ،‬مما يشير إىل وضع التشغيل الفائق‪.‬‬‫	•اضغط عىل زر التشغيل للتصليب‪ .‬إليقاف التصليب قبل إكمال فترة زمنية فاصلة‪ ،‬اضغط عىل زر التشغيل مر ًة أخرى‪.‬‬‫	•للعودة إىل وضع التشغيل القياسي‪ ،‬اضغط مع االستمرار عىل زر الوقت‪ /‬الوضع لمدة ثانيتين ثم حرره‪ .‬يُصدر مصباح الوضع‪ /‬الحالة ضوءًا أخضر وتضيء مصابيح التوقيت الخضراء‪ ،‬مما يشير إىل وضع‬‫التشغيل القياسي‪.‬‬‫وضع السكون‪ :‬ينتقل جهاز التصليب الضوئي إىل وضع السكون بعد ‪ 60‬ثانية من عدم النشاط‪ ،‬ويتضح ذلك من خالل صدور وميض بطيء من مصباح الوضع‪ /‬الحالة‪ .‬يؤدي اإلمساك بالوحدة أو لمسها إىل إيقاظ‬‫جهاز التصليب الضوئي وإعادته تلقائ ًيا إىل آخر إعداد ُمستخدَم‪ .‬لزيادة عمر البطارية إىل أقصى حد‪ ،‬تجنب تحريك جهاز التصليب الضوئي عندما ال يكون قيد االستعمال‪.‬‬‫التنظيف‬‫(‪1‬يرجى التخلص من القفازات الطبية المستخدمة في حاوية نفايات قياسية بعد معالجة كل مريض‪.‬‬‫(‪2‬انظر قسم المعالجة‪.‬‬‫تعليمات استخدام سناد التثبيت‬‫خال من الزيوت‪.‬‬‫(‪1‬يجب تثبيت السناد عىل سطح‬‫ٍ‬‫مستو ٍ‬‫(‪2‬يرجى تنظيف السطح بفركه بالكحول‪.‬‬‫(‪3‬يرجى نزع الجزء الخلفي من الشريط الالصق للسناد‪.‬‬‫(‪4‬يرجى وضع السناد في موضع يمكّن من رفع جهاز التصليب الضوئي ألعىل عند نزعه‪ .‬اضغط بقوة عىل هذا الموضع‪.‬‬‫دليل األوضاع السريع‪:‬‬‫‪299‬‬

‫الطاقة القياسية‬‫‪ 1000‬ميجاوات ‪/‬سم‪2‬‬‫الوضع‬‫الطاقة العالية الزائدة‬‫‪ 1600‬ميجاوات ‪/‬سم‪2‬‬‫يُرفق مع جهاز التصليب الضوئي ‪ 4‬بطاريات ليثيوم فوسفات الحديد قابلة إلعادة الشحن‪.‬‬‫الطاقة الزائدة‬‫‪ 3200‬ميجاوات ‪/‬سم‪2‬‬‫كيفية شحن البطاريات‪:‬‬‫(‪1‬قم بتوصيل الشاحن بمأخذ التيار الكهربائي‪.‬‬‫(‪2‬أدخل البطاريات في الشاحن بحيث يكون الطرف الموجب (‪ )+‬في اتجاه مصابيح المؤشرات عىل الشاحن‪.‬‬‫(‪3‬تشير المصابيح الحمراء إىل أن البطاريات جاهزة لالستخدام‪.‬‬‫(‪4‬يستغرق شحن البطاريات ‪ 3-1‬ساعات‪ .‬اترك البطاريات في الشاحن حتى تكون جاهزة لالستخدام‪.‬‬‫ملحوظة‪ :‬إذا لم يتحول الضوء األحمر الصادر عن الشاحن إىل اللون األخضر عند شحن البطاريات لمدة تزيد عن ثالث ساعات‪ ،‬فقد ينخفض أداء البطارية (تتلف) وال يمكن شحنها‪ .‬جرب بطارية جديدة أو اتصل‬‫بخدمة العمالء في ‪ Ultradent‬لطلب مجموعة جديدة من البطاريات القابلة إلعادة الشحن‪.‬‬‫زر الطاقة‬‫مصابيح ‪ LED‬الخاصة بالوضع‪/‬‬‫التوقيت‬‫كيفية تغيير‪ /‬إدخال البطاريات‪:‬‬‫(‪1‬قم بإزالة الغطاء الخلفي عن طريق تدويرها في عكس اتجاه حركة عقارب الساعة ربع دورة‪.‬‬‫(‪2‬قم بإزالة البطاريات‪.‬‬‫(‪3‬أدخل بطاريات جديدة بإدخال الجانب الموجب (‪ )+‬أوال ً‪.‬‬‫(‪4‬أعد ربط الغطاء الخلفي عن طريق المحاذاة والدفع بلطف أثناء التدوير في اتجاه حركة عقارب الساعة‪ .‬يصدر صوت نقرة من الغطاء عند ربطه بالكامل‪.‬‬‫(‪5‬الوحدة جاهزة لالستخدام‪.‬‬‫إذا لزم األمر‪ ،‬تصرح ‪ Ultradent‬بالبطاريات غير القابلة إلعادة الشحن التالية لجهاز التصليب الضوئي‪:‬‬‫ملحوظة‪ :‬ال تحاول شحن البطاريات غير القابلة إلعادة الشحن‪.‬‬‫أزرار الوقت‬‫خيارات الوقت‬‫‪5‬ث‬‫‪1‬ث‬‫‪2‬ث‬‫‪1‬ث‬‫‪1‬ث‬‫‪2‬ث‬‫لتغيير الوقت‬‫‪4‬ث‬‫‪3‬ث‬‫‪3‬ث فقط‬‫اضغط عىل زر الوقت واتركه بسرعة للتدوير بين خيارات الوقت‪.‬‬‫ضغطا مطوال ً عىل مفتاح الوقت لمدة ثانيتين ثم حرره‪ .‬سينتقل جهاز التصليب الضوئي إىل الوضع التالي‪.‬‬‫اضغط‬‫ً‬‫لتغيير األوضاع‬‫مفاتيح الرموز‬‫مصابيح ‪ LED‬الوامضة‬‫مصابيح ‪ LED‬الصلبة‬‫دليل التصلب السريع‪:‬‬‫)‪Energy** (Joules‬‬‫)‪Total Exposure Time (Seconds‬‬‫)‪Irradiance* (mW/cm2‬‬‫)‪Power* (mW‬‬‫‪Curing Mode‬‬‫‪4.8J/9.7J/14.5J/19.4J‬‬‫‪5/10/15/20‬‬‫‪900‬‬‫‪970‬‬‫‪Standard‬‬‫‪1.6J/3.2J/4.8J/6.5J‬‬‫‪1/2/3/4‬‬‫‪1,500‬‬‫‪1,620‬‬‫‪High Power Plus‬‬‫‪6.8J‬‬‫‪3‬‬‫‪2,100‬‬‫‪2,260‬‬‫‪Xtra‬‬‫ً‬‫منخفضا للغاية‪،‬‬‫بطاريات منخفضة‪ :‬يُصدر جهاز التصليب الضوئي إشارة للمستخدم بأنه قد حان الوقت لتغيير البطاريات عندما يومض مصباح مؤشر البطارية المنخفضة باللون األحمر‪ .‬إذا أصبح شحن البطارية‬‫يصدر تحذير صوتي يتألف من ‪ 3‬إشارات صوتية وال يسمح جهاز التصليب الضوئي بإجراء مزيد من عمليات التشغيل حتى تتم إعادة شحن البطاريات أو إدخال بطاريات جديدة‪( .‬انظر دليل التحذير السريع)‬‫دليل تحذير سريع‪:‬‬‫تحذير من مستوى الطاقة‬‫استبدل البطاريات‬‫	•البطارية منخفضة‪ :‬وميض بطيء‬‫	•التوقف عن العمل‪ 3 :‬صافرات‪ ،‬وميض‬‫	•الحيلولة دون التشغيل‬‫تحذير من درجة الحرارة‬‫السماح بالتبريد‬‫	•‪ 3‬صافرات‬‫	•وميض بطيء‬‫	•الحيلولة دون التشغيل‬‫تحذير بالخدمة‬‫اتصل بخدمة العمالء لإلصالح‬‫	•ال يوجد صوت‬‫	•وميض‪ ،‬كل ثانيتين‬‫	•يسمح بالتشغيل‬‫(‪Tenergy Propel Photo Lithium1‬‬‫(‪Titanium Innovations CR123A2‬‬‫(‪Energizer® 1233‬‬‫(‪Duracell® Ultra CR123A4‬‬‫(‪SureFire® SF123A5‬‬‫(‪Panasonic® CR123A6‬‬‫تحذير لمصباح ‪LED‬‬‫بشكل عام‪ ،‬يجب‬‫وقت الشحن وعمر البطارية‪ :‬يعتمد عمر البطارية المشحونة بالكامل في جهاز التصليب الضوئي عىل الفترة الزمنية الفاصلة للوضع‪ /‬الوقت ونوع البطارية ومقدار االستخدام وكفاءة مصباح ‪.LED‬‬‫ٍ‬‫أن تستمر البطاريات القابلة إلعادة الشحن بين أسبوع وأسبوعين‪ .‬قد تستمر البطاريات غير القابلة إلعادة الشحن لفترة أطول من ‪ 3-2‬مرات‪.‬‬‫	•الفترة الزمنية الموصى بها إلعادة الشحن‪ :‬عندما يُضيء مؤشر البطارية المنخفض‪ ،‬أو كل أسبوع أو أسبوعين تقري ًبا‪ ،‬حسب االستخدام‪.‬‬‫	•البطاريات اإلضافية‪ :‬يُرفق مع جهاز التصليب الضوئي ‪ 4‬بطاريات قابلة إلعادة الشحن‪ .‬نوصي ً‬‫أيضا باالحتفاظ بمجموعة احتياطية من بطاريات ‪ CR123A‬غير القابلة إلعادة الشحن في حالة حدوث مشكلة أو‬‫فقدان البطاريات القابلة إلعادة الشحن‪.‬‬‫	•عمر البطارية المتوقع‪ :‬يمكن إعادة شحن بطاريات ليثيوم فوسفات الحديد القابلة إلعادة الشحن حوالي ‪ 1000‬إىل ‪ 2000‬مرة‪ .‬استنادًا إىل االستخدام العادي والرعاية المناسبة‪ ،‬يجب أن تستمر البطاريات مدة‬‫تصل إىل خمس سنوات‪ ،‬ولكن يجب استبدالها عند اللزوم‪.‬‬‫الضمان‬‫تضمن ‪ Ultradent‬بموجب هذا الضمان أن تتوافق هذه األداة‪ ،‬لمدة ‪ 5‬سنوات*‪ ،‬من جميع النواحي الجوهرية مع المواصفات‪ ،‬وأن تكون خالية من أي عيوب في المواد‪ /‬أو جودة العمل‪ ،‬عىل النحو المنصوص‬‫عليه في وثائق ‪ Ultradent‬المرفقة مع المنتج‪ .‬وينطبق هذا الضمان فقط عىل المشتري األصلي وهو غير قابل للتحويل‪ .‬وتتم إعادة جميع المنتجات المعيبة إىل ‪ .Ultradent‬وال توجد مكونات خدمة مستخدم‬‫لنظام التصليب الضوئي لجهاز فالو كوردليس‪ .‬ويؤدي العبث بجهاز التصليب الضوئي فالو كوردليس إىل إلغاء الضمان‪.‬‬‫وال يشمل ضمان جهاز التصليب الضوئي فالو كوردليس األضرار التي يسببها العمالء‪ .‬فمثال؛ في حالة إساءة استخدام فالو كوردليس أو سقوطه وكسر العدسة‪ ،‬يكون العميل مسؤوال ً عن دفع تكاليف أي‬‫إصالحات ضرورية‪.‬‬‫اتصل بخدمة العمالء لإلصالح‬‫	•‪ 3‬صافرات مستمرة‬‫	•وميض واحد سريع‬‫	•الحيلولة دون التشغيل‬‫*مع إيصال مبيعات يوضح تاريخ البيع لطبيب األسنان‪.‬‬‫المعالجة‬‫بعد كل استخدام‪ ،‬قم بترطيب الشاش أو القماش الناعم باستخدام مطهر السطح المعتمد ومسح السطح والعدسة‪.‬‬‫الصيانة‬‫اإلصالح‬‫اإلصالحات التي يُجريها المستخدم‬‫(‪1‬تحقق بشكل روتيني من العدسة للتأكد من عدم راتنجات األسنان المتصلبة‪ .‬وإذا لزم األمر‪ ،‬فاستخدم أداة أسنان ليس بها ماس إلزالة أي راتنج ملتصق بعناية‪.‬‬‫ً‬‫اختالفا كبيرًا‪ ،‬وهي مصممة لتقديم نصائح إرشادية خاصة بالضوء والعدسات‪ .‬توصي ‪ Ultradent‬بفحص المأخذ بشكل روتيني في وضع التشغيل القياسي‪ .‬مالحظة‪ :‬يتم تحريف الناتج‬‫(‪2‬تختلف عدادات الضوء‬‫الرقمي الحقيقي بسبب عدم دقة عدادات الضوء الشائعة وحزمة الصمام الثنائي الباعث للضوء المخصصة في جهاز التصليب الضوئي‪.‬‬‫إصالحات الشركة المصنعة‬‫(‪1‬ال يتم إجراء اإلصالحات إال من قبل موظفي الخدمة المعتمدين‪ .‬وتوفر ‪ Ultradent‬موظفي خدمة معتمدين إلجراء اإلصالحات‪.‬‬‫دائما عىل نزع البطاريات من جهاز التصليب الضوئي والشاحن‪ .‬لف البطاريات والشاحن والمحول وجهاز التصليب الضوئي بشكل منفصل في‬‫(‪2‬عند إرسال وحدات إلصالحها أو صيانتها أو معايرتها‪ ،‬احرص ً‬‫صندوق اإلرجاع‪.‬‬‫(‪3‬اشحن البطاريات وف ًقا للوائح المحلية‪.‬‬‫صيانة البطارية‬‫شحن وتغيير البطاريات‬‫‪300‬‬‫المنظفات المقبولة‪:‬‬‫	•بخاخ مطهر ‪( Lysol Brand III‬موصى به)‬‫	•كحول األيزوبروبيل‬‫	•المنظفات القائمة عىل الكحول اإليثيلي‬‫	• ُمركِّز ‪( *®Lysol‬القائم عىل الكحول فقط)‬‫المنظفات غير المقبولة ‪ -‬تجنب استخدامها‪:‬‬‫	•المنظفات القلوية القوية من أي نوع‪ ،‬بما في ذلك صابون اليد وصابون تنظيف األطباق‬‫	•المنظفات القائمة عىل التبييض (مثل ‪ *™Clorox‬و‪)*™Sterilox‬‬‫	•المنظفات القائمة عىل بيروكسيد الهيدروجين‬‫	•المنظفات الكاشطة (مثل ‪)*™Comet Cleanser‬‬‫	•المنظفات القائمة عىل األسيتون أو الهيدروكربون‬‫	•ميثيل إيثيل كيتون (‪)MEK‬‬‫	•‪*®Birex‬‬‫	•‪Gluteraldehyde‬‬‫	•المنظفات القائمة عىل أمالح رباعي األمونيوم كلوريدمحلول أو مناديل ‪*™Cavicide1‬‬‫	•منتجات ‪( *™Cavicide‬غير المبيضة)**‬‫* عالمة تجارية لشركة غير ‪Ultradent‬‬‫** إذا استخدمت‪ ،‬فقد تؤدي إىل أن يبهت اللون‬‫شاحن البطارية‪:‬‬‫‪301‬‬

‫المعلومات‪ /‬البيانات الفنية‬‫تماما قبل استئناف االستخدام‪.‬‬‫إذا كان التنظيف ضروري ًا‪ ،‬فافصل الشاحن‪ ،‬ثم رطب قطعة قماش بكحول األيزوبروبيل‪ ،‬وامسح سطح الشاحن أو البطاريات برفق‪ .‬اترك الشاحن يجف‬‫ً‬‫نطاقات تردد طول الموجة الفعالة المركبة‪:‬‬‫تنظيف واقي المصباح‪:‬‬‫قم بتطهير واقي مصباح فالو كوردليس باستخدام أي مطهر أسطح‪ .‬تجنب استخدام األوتوكالف‪.‬‬‫التخزين والتخلص‬‫دائما عىل نزع البطاريات‪ .‬إذا تُركت البطاريات في الوحدة لفترات زمنية طويلة دون إعادة شحنها‪ ،‬فقد تتلف أو تصبح‬‫في حالة تخزين جهاز التصليب الضوئي لفترات تزيد عن أسبوعين‪ ،‬أو تعبئته للسفر‪ ،‬احرص ً‬‫غير قابلة للشحن‪ .‬تجنب تخزين البطاريات في درجات حرارة تزيد عن ‪ 60‬درجة مئوية (‪ 140‬درجة فهرنهايت) أو في ضوء الشمس المباشر‪.‬‬‫تخزين ونقل جهاز التصليب الضوئي‪:‬‬‫	•درجة الحرارة‪ +10 :‬درجة مئوية إىل ‪ +40‬درجة مئوية (‪ +50‬درجة فهرنهايت إىل ‪ +104‬درجات فهرنهايت)‬‫	•الرطوبة النسبية‪ 10% :‬إىل ‪95%‬‬‫	•الضغط المحيط‪ 500 :‬هيكتوباسكال إىل ‪ 1060‬هيكتوباسكال‬‫دئازلا‬‫عند التخلص من النفايات اإللكترونية‪( ،‬أي أجهزة التصليب الضوئي وأجهزة الشحن والبطاريات ووحدات اإلمداد بالطاقة)‪ ،‬يرجى اتباع اإلرشادات المحلية للنفايات وإعادة التدوير‪.‬‬‫ةيلاعلا ةقاطلا‬‫يسايق‬‫االعتبارات الفنية‬‫الملحقات‬‫العنصر‬‫القفازات الطبية ‪VALO‬‬‫واقي مصباح فالو كوردليس‬‫شاحن ‪VALO‬‬‫معلومات ‪CE‬‬‫‪MDSS GmbH‬‬‫‪Schiffgraben 41‬‬‫‪30175 Hanover‬‬‫‪Germany‬‬‫جهة التصنيع‪:‬‬‫‪.TIDI Products, LLC‬‬‫‪Enterprise Drive 570‬‬‫‪Neenah, WI 54956‬‬‫ُصنع في الواليات المتحدة األمريكية‬‫توزيع‪:‬‬‫‪Ultradent Products Inc‬‬‫‪West Ultradent Drive (10200 505‬‬‫‪)South‬‬‫‪South Jordan, UT 84095‬‬‫‪USA‬‬‫‪:‬‬‫)رتمونان( ‪WAVELENGTH‬‬‫السمات‬‫المعلومات‪ /‬المواصفات‬‫العدسة‬‫قطر ‪ 11.7‬مم‬‫نطاق الطول الموجي‬‫	•نطاق الطول الموجي المتاح‪ 515 - 385 :‬نانومتر‬‫	•الحد األقصى لألطوال الموجية‪ 415 - 395 :‬نانومتر و‪ 480 - 440‬نانومتر‬‫يختلف االنبعاث اإلشعاعي بنا ًء عىل قدرة الجهاز‬‫وطريقة القياس وموضع الضوء‪.‬‬‫† يجب أن يستخدم مقياس ضوء تصليب الحشوات‬‫ومحلل الطيف ‪ MARC‬كمرجع فقط نظرًا لوجود‬‫فتحات أصغر من مصابيح التصليب ‪.VALO‬‬‫* يجب أن يُستخدم مقياس ضوء تصليب الحشوات‬‫كمرجع فقط بسبب قيود الطاقة واالستجابة‬‫الطيفية‪.‬‬‫‡ يتطابق االنبعاث اإلشعاعي مع المعيار ‪ISO‬‬‫‪ 10650‬عند قياسه بمحلل طيف يعمل بوحدة‬‫الجيجاهرتز‪.‬‬‫‪Nominal Radiant Exitance Comparison Chart‬‬‫بطاريات ‪VALO‬‬‫†* ‪Demetron‬‬‫‪.L.E.D‬‬‫المقياس‬‫أداة القياس‬‫جدول كثافة الضوء‬‫فتحة العداد‬‫‪ 7‬ملم‬‫الطاقة القياسية‬‫(‪)±10%‬‬‫‪ 1000‬ميجاوات‪/‬‬‫سم‪²‬‬‫الطاقة العالية الزائدة‬‫(‪)±10٪‬‬‫‪ 1600‬ميجاوات‪/‬‬‫سم‪²‬‬‫التشغيل الفائق‬‫(‪)10٪ ±‬‬‫‪302‬‬‫يجوملا لوطلا قاطن‬‫لاعفلا يولعلا‬‫رتمونان ‪440-480‬‬‫ةلاعفلا ةيلفسلا‬‫يجوملا لوطلا قاطن‬‫رتمونان ‪395-415‬‬‫)رتمونان‪/‬طاواغيم( ةيفيطلا ةيعاعشإلا ةقاطلا‬‫ةلاعفلا ةبكرملا ةجلاعملا يجوملا لوطلا تاقاطن‬‫)رتمونان ‪: 385-515‬مادختسالل لباقلا يجوملا لوطلا قاطن(‬‫† تسجيلة الفهرسة‬‫المقروءة آل ًيا‬‫(‪)MARC‬‬‫محلل الطيف‬‫االنبعاث‬‫‪ 3.9‬ملم‬‫‡ محلل الطيف بوحدة الجيجاهيرتز‬‫إجمالي الطاقة‬‫‪ 15‬ملم‬‫‪ 15‬ملم‬‫‪ 900‬ميجاوات‬‫‪ /‬سم‪²‬‬‫‪ 970‬ميجاوات‬‫‪ 1800‬ميجاوات‪/‬‬‫سم‪2‬‬‫‪ 1500‬ميجاوات‬‫‪ /‬سم‪²‬‬‫‪ 1615‬ميجاوات‬‫‪ 3200‬ميجاوات ‪/‬‬
‫سم‪)20٪ -/+( ²‬‬‫‪ 2100‬ميجاوات‬‫‪ /‬سم‪²‬‬‫‪ 2260‬ميجاوات‬‫جهاز التصليب‬‫الضوئي فالو جراند‬‫التصنيفات‪( 1-IEC 60601 :‬السالمة)‪( 2-1-IEC 60601 ،‬التوافق الكهرومغناطيسي)‬‫الوزن‪:‬‬‫غراما)‬‫	•مع البطاريات‪ 6 :‬أوقيات‪190( .‬‬‫ً‬‫	•بدون البطاريات‪ 5 :‬أوقيات‪ 150( .‬غراما)‬‫األبعاد‪ )1.06 × 1.28 × 8( :‬بوصة‪ )27 × 32.5 × 203( ،‬مم‬‫وحدة اإلمداد بالطاقة‬‫للشاحن‬‫الخرج ‪ 12 -‬فولت تيار مباشر في ‪ 500‬مللي أمبير‬‫الدخل ‪ 100 -‬فولت تيار متردد إىل ‪ 240‬فولت تيار متردد‬‫‪ Ultradent P/N 5930‬وحدة إمداد الطاقة لجهاز ‪ VALO‬بمقابس عالمية‬‫التصنيفات‪( 1-IEC 60601 :‬السالمة)‬‫طول السلك ‪ 6 -‬أقدام (‪ 1.8‬متر)‬‫وحدة إمداد الطاقة لجهاز ‪ VALO‬هي وحدة إمداد طاقة ذات درجة طبية‬‫ثانية وتوفر االنعزال عن التيار الكهربائي الرئيسي‬‫‪303‬‬

‫السمات‬‫المعلومات‪ /‬المواصفات‬‫شاحن ‪VALO‬‬‫شاحن البطارية الذكي ‪ VALO 3,6‬فولت تيار مستمر للبطاريات المصنوعة من فوسفات حديد ليثيوم‪:‬‬‫	•إيقاف التشغيل التلقائي عند شحن البطارية بالكامل‬‫	•الكشف التلقائي عن البطاريات المعيبة‬‫	•وسائل الحماية‪ :‬الحرارية‪ ،‬الشحن الزائد‪ ،‬ماس كهربائي‪ ،‬عكس القطبية‬‫	⸰الصمام الثنائي الباعث للضوء يُصدر ضوءًا أحمر – جار الشحن‬‫	⸰الصمام الثنائي الباعث للضوء يُصدر ضوءًا أخضر ‪ -‬البطارية فارغة أو مشحونة بالكامل‬‫	⸰الصمام الثنائي الباعث للضوء ال يُصدر ضوءًا – ماس كهربائي‬‫	•مدة الشحن‪ 3 - 1 :‬ساعات‬‫التصنيف‪CE, WEEE :‬‬‫بطاريات ‪VALO‬‬‫القابلة إلعادة الشحن‪ :‬بطارية ليثيوم فوسفات الحديد اآلمنة كيميائ ًيا (‪LiFePO4) RCR123A‬‬‫	•فولطية التشغيل‪ 3،2 :‬فولت تيار مستمر‬‫التصنيفات‪UL, CE, RoHS, WEEE :‬‬‫ظروف التشغيل‬‫درجة الحرارة‪ +10 :‬درجة مئوية إىل ‪ +32‬درجة مئوية (‪ +50‬درجة فهرنهايت إىل ‪ +90‬درجة فهرنهايت)‬‫الرطوبة النسبية‪ 10٪ :‬إىل ‪95٪‬‬‫الضغط المحيط‪ 700 :‬هيكتوباسكال إىل ‪ 1060‬هيكتوباسكال‬‫دورة التشغيل‬‫تم تصميم جهاز التصليب الضوئي للتشغيل عىل المدى القصير‪ .‬في درجة الحرارة المحيطة القصوى (‪ 32‬درجة مئوية) لمدة دقيقة تشغيل واحدة تتم إعادة المعالجة‬‫بشكل دوري‪ ،‬إيقاف التشغيل لمدة ‪ 30‬دقيقة (فترة التبريد)‪.‬‬‫معلومات متنوعة‬‫التوجيه وإعالن الشركة المصنعة بشأن االنبعاثات الكهرومغناطيسية‬‫جهاز التصليب الضوئي مخصص لالستخدام في البيئة الكهرومغناطيسية المحددة أدناه‪ .‬ويجب عىل العميل أو المستخدم التأكد من استخدامه في مثل هذه البيئة‪.‬‬‫اختبار االنبعاثات‬‫االمتثال‬‫البيئة الكهرومغناطيسية ‪ -‬التوجيه‬‫انبعاثات الترددات الالسلكية‬‫‪CISPR 11‬‬‫المجموعة ‪1‬‬‫يستخدم جهاز التصليب الضوئي البطاريات وال يتأثر بالتداخل الكهرومغناطيسي أو الترددات الالسلكية أو منع‬‫اندفاع التيار‪.‬‬‫انبعاثات الترددات الالسلكية‬‫‪CISPR 11‬‬‫الفئة ب‬‫يستخدم جهاز التصليب الضوئي الطاقة الكهربائية والكهرومغناطيسية فقط في وظائفها الداخلية‪ .‬ولذلك‪ ،‬فإن‬‫أي انبعاثات ترددات السلكية منخفضة جدًا وال يرجح أن تسبب تداخال ً في المعدات اإللكترونية القريبة‪.‬‬‫االنبعاثات المتناغمة‬‫‪2-3-IEC 61000‬‬‫غير متوفر‬‫المعيار ‪ 3-3-IEC 61000‬لتقلبات‬‫الجهد‪ /‬انبعاثات الوميض‬‫غير متوفر‬‫ال ينطبق اختبار االنبعاث المتزامن وتذبذب التيار الكهربائي عىل جهاز التصليب الضوئي ألنه يعمل بالبطارية‪.‬‬‫جهاز التصليب الضوئي مالئم لالستخدام في جميع المؤسسات‪ ،‬بما في ذلك المنشآت المحلية والمرتبطة‬‫ارتباطا مباشرًا بشبكة إمداد الطاقة العامة منخفضة الجهد التي تمد المباني بالطاقة الالزمة لالستخدام المنزلي‪.‬‬‫ً‬‫استكشاف األخطاء وإصالحها‬‫إذا كانت الحلول المقترحة أدناه ال تفي بتصحيح المشكلة‪ ،‬فيرجى االتصال بشركة ‪ Ultradent‬عىل الرقم ‪ .800.552.5512‬خارج الواليات المتحدة‪ ،‬اتصل بموزع ‪ Ultradent‬أو تاجر األغراض‬‫المتعلقة باألسنان‪.‬‬‫المشكلة‬‫الحلول الممكنة‬‫المصباح ال يُضيء‬‫(‪1‬اضغط عىل مفتاح تغيير الوقت‪ /‬الوضع أو مفتاح الطاقة للخروج من وضع توفير الطاقة‪.‬‬‫(‪2‬افحص مؤشر البطارية المنخفضة باللون األحمر لمعرفة حالة شحن البطارية‪.‬‬‫(‪3‬تأكد من إدخال البطاريات الجديدة بشكل صحيح في الوحدة‪.‬‬‫(‪4‬إذا أومضت مؤشرات التحذير باللونين األحمر واألصفر‪ ،‬فهذا يعني أن جهاز التصليب الضوئي قد وصل إىل حد سالمة درجة الحرارة‬‫الداخلية‪ .‬اترك جهاز التصليب الضوئي يبرد لمدة ‪ 10‬دقائق أو استخدم منشفة رطبة باردة لتبريد الوحدة بسرعة‪.‬‬‫(‪5‬إذا كان مؤشر التحذير باللون األحمر يومض ويصدر صوتًا مستمرًا‪ ،‬فاتصل بخدمة عمالء ‪ Ultradent‬لإلصالح‪.‬‬‫المصباح ال يُضيء للوقت المطلوب‬‫(‪1‬تحقق من وضع وأضواء التوقيت إلدخال الوقت الصحيح‪.‬‬‫(‪2‬افحص مؤشر البطارية المنخفضة لمعرفة حالة شحن البطارية‪.‬‬‫(‪3‬تأكد من إدخال البطاريات الجديدة بشكل صحيح في الوحدة‪.‬‬‫الضوء ال يقوم بتصليب الراتنجات‬‫بشكل صحيح‬‫(‪1‬افحص العدسة للوقوف عىل وجود الراتنجات‪ /‬المركبات المعالجة‪.‬‬‫(‪2‬باستخدام واقي العينين كهرماني اللون الذي يحمي من األشعة فوق البنفسجية‪ ،‬تحقق من عمل مصابيح الصمام الثنائي الباعث للضوء‬‫(‪.)LED‬‬‫(‪3‬تحقق من مستوى الطاقة بمقياس الضوء‪ .‬في حالة استخدام مقياس ضوء‪ ،‬توصي ‪ Ultradent‬بفحص جهاز التصليب الضوئي في وضع‬‫التشغيل القياسي‪.‬‬‫ملحوظة‪ :‬سيتم تحريف الناتج الرقمي الحقيقي بسبب عدم دقة وحدات قياس الضوء الشائعة وحزمة ‪ LED‬المخصصة التي يستخدمها جهاز‬‫ً‬‫اختالفا كبيرًا‪ ،‬وهي مصممة لتقديم نصائح إرشادية خاصة بالضوء والعدسات‪.‬‬‫التصليب الضوئي‪ .‬تختلف عدادات الضوء‬‫(‪4‬تحقق من تاريخ انتهاء الصالحية فيما يتعلق بتصليب الراتنج‪.‬‬‫(‪5‬تأكد من اتباع األسلوب المناسب وف ًقا لتوصيات الشركة المصنعة‪.‬‬‫البطاريات ال تشحن‬‫(‪1‬تأكد من إدخال البطاريات في الشاحن في االتجاه الصحيح واترك البطاريات لشحنها لمدة ‪ 3-1‬ساعات‪.‬‬‫(‪2‬إذا لم يتغير ضوء المصابيح الحمراء عىل الشاحن إىل اللون األخضر‪ ،‬فاتصل بخدمة عمالء ‪ Ultradent‬لطلب بطاريات بديلة و‪ /‬أو شاحن‬‫بديل‪.‬‬‫(‪3‬إذا لم يكن هناك ضوء أخضر أو أحمر عىل الشاحن‪ ،‬فاتصل بخدمة عمالء ‪ Ultradent‬لطلب أو استبدال الشاحن و ‪ /‬أو محول التيار‬‫المتردد‪.‬‬‫الشاحن ال يشحن البطاريات‬‫(‪1‬تأكد من توصيل الشاحن وأن محول التيار المتردد متصل بمأخذ طاقة يعمل‪.‬‬‫(‪2‬إذا كانت المصابيح الخضراء أو الحمراء عىل الشاحن غير مرئية‪ ،‬فاتصل بخدمة عمالء ‪ Ultradent‬للحصول عىل شاحن جديد و‪ /‬أو محول‬‫تيار متردد‪.‬‬‫ال يمكن تغيير الفواصل الزمنية أو الوضع‬‫ضغطا مطوال ً عىل مفتاحي الوقت‪ /‬الوضع والطاقة إىل أن تشير سلسلة من األصوات إىل أن ضوء المعالجة مفتوح‪.‬‬‫(‪1‬اضغط‬‫ً‬‫التوجيه وإعالن الشركة المصنعة للحصانة الكهرومغناطيسية‬‫جهاز التصليب الضوئي مخصص لالستخدام في البيئة الكهرومغناطيسية المحددة أدناه‪ .‬ويجب عىل العميل أو المستخدم التأكد من استخدامه في مثل هذه البيئة‪.‬‬‫اختبار الحصانة‬‫مستوى االختبار ‪IEC 60601‬‬‫درجة االمتثال‬‫إرشادات البيئة الكهرومغناطيسية‬‫التفريغ الكهربائي الساكن‬‫‪ ±8‬كيلو فولت لالتصال‬‫‪ ±15‬كيلو فولت للهواء‬‫‪ ±8‬كيلو فولت لالتصال‬‫‪ ±15‬كيلو فولت للهواء‬‫وينبغي أن تقتصر البيئة المادية عىل ما يلي‪:‬‬‫(‪1‬رمز الحماية من المواد الدخيلة‪IP20 :‬‬‫(‪2‬تجنب غمره في سائل‪.‬‬‫(‪3‬تجنب استخدامه حول الغازات القابلة لالشتعال‪ .‬الوحدة‬‫ليست ‪ APG‬وليست ‪.AP‬‬‫(‪4‬نطاق تخزين الرطوبة‪95% - 10% :‬‬‫(‪5‬نطاق تراوح درجات حرارة التخزين‪ 10 :‬درجات مئوية‬‫‪ 40 -‬درجة مئوية‬‫الدفق الكهربائي السريع‪/‬االنفجار‬‫‪4-4-IEC 61000‬‬‫‪ 2 ±‬كيلو فولت لخطوط إمدادات‬‫الطاقة‬‫‪ 1 ±‬كيلو فولت لخطوط الدخل‪ /‬الخرج‬‫‪ 2 ±‬كيلو فولت لخطوط إمدادات‬‫الطاقة‬‫مالحظة ‪ :1‬ال يحتوي جهاز التصليب‬‫الضوئي عىل منافذ ‪I/0‬‬‫جهاز التصليب الضوئي يعمل بالبطارية وال يمكنه االتصال‬‫بالتيار المتردد الرئيسي‪.‬‬‫الزيادة السريعة‬‫‪5-4-IEC 61000‬‬‫‪ 1 ±‬كيلو فولت خط إىل خط‬‫‪ 2 ±‬كيلو فولت خط إىل أرضي‬‫‪ 1 ±‬كيلو فولت خط إىل خط‬‫‪ 2 ±‬كيلو فولت خط إىل أرضي‬‫الفولتية‪ ،‬االنخفاضات‪ ،‬والماس‪،‬‬‫واالنقطاعات واالختالفات في خطوط‬‫مدخالت إمدادات الطاقة‬‫<‪U 5%‬‬‫(> تراجع ‪ 95٪‬في ‪ U‬لمدة ‪ 0.5‬دورة)‬‫<‪U 5%‬‬‫(> تراجع ‪ 95٪‬في ‪ U‬لمدة ‪ 0.5‬دورة)‬‫‪U 40%‬‬‫(تراجع ‪ 60%‬في ‪ U‬لمدة ‪ 5‬دورات)‬‫‪U 40%‬‬‫(تراجع ‪ 60%‬في ‪ U‬لمدة ‪ 5‬دورات)‬‫‪U 70%‬‬‫(تراجع ‪ 30%‬في ‪ U‬لمدة ‪ 25‬دورة)‬‫‪U 70%‬‬‫(تراجع ‪ 30%‬في ‪ U‬لمدة ‪ 25‬دورة)‬‫<‪U 5%‬‬‫ثوان)‬‫(> تراجع ‪ 95٪‬في ‪ U‬لمدة ‪5‬‬‫ٍ‬‫<‪U 5%‬‬‫ثوان)‬‫(> تراجع ‪ 95٪‬في ‪ U‬لمدة ‪5‬‬‫ٍ‬‫مالحظة ‪ :2‬التعافي الذاتي‬‫في حالة انخفاض الجهد الكهربائي لجهاز التصليب الضوئي‬‫إىل ‪ 4‬فولت تيار مستمر‪ ،‬لن تسمح الوحدة بالتشغيل‪.‬‬‫وسيتم إيقاف تشغيل جهاز التصليب الضوئي فالو كوردليس‪.‬‬‫وعند إدخال بطاريات جديدة واستعادة مستويات الطاقة‬‫المناسبة‪ ،‬تتم إعادة تشغيل جهاز التصليب الضوئي ويعود‬‫إىل الحالة نفسها قبل انقطاع التيار‪ .‬يستعيد جهاز التصليب‬‫الضوئي حالته ذات ًيا في حالة انقطاع التيار‪.‬‬‫‪ 30‬أ‪/‬م‬‫‪ 30‬أ‪/‬م‬‫يجب أن تكون المجاالت المغناطيسية لتردد الطاقة‬‫بمستويات تتميز بموقع نموذجي في بيئة نموذجية أو سكنية‬‫أو رعاية صحية منزلية أو تجارية أو خاصة بالمستشفى‬‫أو عسكرية‪.‬‬‫‪2-4-IEC 61000‬‬‫‪11-4-IEC 61000‬‬‫تردد الطاقة‬‫(‪ 60/50‬هرتز) المجال المغناطيسي‬‫‪8-4-IEC 61000‬‬‫نظرًا ألن جهاز التصليب الضوئي يعمل بالبطارية‪ ،‬فإنه‬‫ال يخضع لنقل الكهرباء أو العواصف أو االنخفاضات في‬‫الجهد أو انقطاع التيار أو االنقطاعات أو االختالفات في‬‫طاقة التيار المتردد‪.‬‬‫قد يخضع شاحن البطاريات الملحق لما ورد أعاله‪ ،‬لكنه‬‫منفصل عن تشغيل جهاز التصليب الضوئي فالو كوردليس‬‫حاسما له‪.‬‬‫وليس‬‫ً‬‫مالحظة ‪ :1‬جهاز التصليب الضوئي غير مزود بأي منافذ أو أي خطوط دخل‪ /‬خرج يمكن الوصول إليها‪.‬‬‫‪304‬‬‫‪305‬‬

‫التوجيه وإعالن الشركة المصنعة لمسافة الفصل الموصى بها بين معدات االتصاالت الالسلكية المحمولة والمتحركة وجهاز التصليب الضوئي فالو كوردليس‬‫التوجيه وإعالن الشركة المصنعة للحصانة الكهرومغناطيسية ألنظمة الدعم غير الحيوية‬‫جهاز التصليب الضوئي مخصص لالستخدام في البيئة الكهرومغناطيسية المحددة أدناه‪ .‬ويجب عىل العميل أو المستخدم التأكد من استخدامه في مثل هذه البيئة‪.‬‬‫اختبار الحصانة‬‫مستوى االختبار ‪IEC‬‬‫‪60601‬‬‫درجة االمتثال‬‫التردد الالسلكي للتوصيل‬‫‪Vrms 3‬‬‫‪Vrms 3‬‬‫‪6-4-IEC 61000‬‬‫‪ 150‬كيلو هرتز إىل ‪80‬‬‫ميجا هرتز‬‫‪ 150‬كيلو هرتز إىل ‪80‬‬‫ميجا هرتز‬‫الترددات الالسلكية‬‫المنبعثة‬‫‪ 3‬فولت ‪ /‬م‬‫‪ 3‬فولت ‪ /‬م‬‫‪3-4-IEC 61000‬‬‫‪ 80‬ميجا هرتز إىل ‪2.5‬‬‫جيجا هرتز‬‫‪ 80‬ميجا هرتز إىل ‪2.5‬‬‫جيجا هرتز‬‫جهاز التصليب الضوئي مخصص لالستخدام في بيئة كهرومغناطيسية يتم فيها التحكم في اضطرابات الترددات الالسلكية المرسلة‪ .‬يمكن لمستخدم جهاز التصليب الضوئي المساعدة في‬‫منع التداخل الكهرومغناطيسي من خالل الحفاظ عىل مسافة دنيا بين معدات االتصاالت الالسلكية المحمولة والمتحركة (أجهزة اإلرسال) وجهاز التصليب الضوئي عىل النحو الموصى به‬‫أدناه‪ ،‬وف ًقا لطاقة خرج معدات االتصاالت القصوى‪.‬‬‫مسافة الفصل وف ًقا لتردد المرسل‬‫(متر)\‬‫إرشادات البيئة الكهرومغناطيسية‬‫يجب عدم استخدام معدات االتصاالت الالسلكية المحمولة بالقرب من أي جزء من جهاز التصليب‬‫الضوئي‪ ،‬بما في ذلك الكابالت‪ ،‬من مسافة الفصل الموصى بها المحسوبة من المعادلة المنطبقة‬‫المرسل‪.‬‬‫عىل تردد ُ‬‫مسافة الفصل الموصى بها‪:‬‬‫‪ 80‬ميجا هرتز إىل ‪ 800‬ميجا هرتز‬‫تصنيف أقصى خرج للطاقة‬‫لجهاز اإلرسال‬‫(‪ P‬بالوات)‬‫كيلوهرتز ‪ 80 -‬ميجاهرتز ‪150‬‬‫ميجاهرتز ‪ 800 -‬ميجاهرتز ‪80‬‬‫‪ 800‬ميجاهرتز – ‪ 2.5‬جيجاهرتز‬‫‪0.01‬‬‫‪ 0.12‬متر‬‫‪ 0.035‬متر‬‫‪ 0.07‬متر‬‫‪0.1‬‬‫‪ 0.37‬متر‬‫‪ 0.11‬متر‬‫‪ 0.22‬متر‬‫‪1‬‬‫‪ 1.7‬متر‬‫‪ 0.35‬متر‬‫‪ 0.7‬متر‬‫‪ 800‬ميجا هرتز إىل ‪ 2.5‬جيجا هرتز‬‫‪10‬‬‫‪ 3.7‬متر‬‫‪ 1.11‬متر‬‫‪ 2.22‬متر‬‫لمصنِّع المرسل‪ ،‬و‪ d‬هي‬‫‪ P‬هي الحد األقصى لمعدل قدرة خرج المرسل بوحدات الوات (‪ )W‬وف ًقا ُ‬‫مسافة المباعدة الموصى بها باألمتار (‪.)m‬‬‫‪100‬‬‫‪ 11.7‬متر‬‫‪ 3.5‬متر‬‫‪ 7.0‬متر‬‫يجب أن تكون شدة المجال الكهربائي من مرسالت التردد الالسلكي الثابتة‪ ،‬كما تحددها دراسة الموقع‬‫الكهرومغناطيسيأ أقل من درجة االمتثال في كل نطاق ترددب‪.‬‬‫قد يحدث تداخل بالقرب من المعدات التي تحمل الرمز التالي‪:‬‬‫تم اختبار جهاز التصليب الضوئي وف ًقا للمعيار ‪ 2014 :2-1-IEC 60601‬وتم تمريره تحت شدة المجال المشع من ‪ 10‬فولت ‪ /‬متر بين ‪ 80‬ميجاهرتز إىل ‪ 2.5‬جيجاهرتز‪ .‬تتطابق قيمة‬‫‪ 3Vrms‬مع ‪ V1‬والقيمة ‪ 10‬ف‪ /‬م تقابل ‪ E1‬في المعادالت أعاله‪.‬‬‫بالنسبة للمرسالت المصنفة عند قدرة خرج قصوى خالف المذكورة أعاله‪ ،‬يمكن تقدير مسافة الفصل الموصى بها ‪ d‬باألمتار (‪ )m‬باستخدام المعادلة المنطبقة عىل تردد المرسل حيث ‪P‬‬‫هي أقصى قدرة خرج للمرسل بالوات (‪ )W‬وف ًقا للجهة المصنّعة للمرسل‪.‬‬‫مالحظة ‪ 1‬عند ‪ 80‬ميجاهيرتز و‪ 800‬ميجاهيرتز‪ ،‬تُطبق مسافة الفصل لنطاق التردد األعىل‪.‬‬‫مالحظة ‪ 2‬قد ال تنطبق هذه اإلرشادات في جميع الحاالت‪ .‬ويتأثر االنتشار الكهرومغناطيسي باالستيعاب واالنعكاس من الهياكل واألشياء واألشخاص‪.‬‬‫مالحظة ‪ 1‬يُستخدم نطاق التردد األعىل عند ‪ 80‬ميجاهرتز و‪ 800‬ميجاهرتز‪.‬‬‫مالحظة ‪ 2‬قد ال تنطبق هذه اإلرشادات في جميع الحاالت‪ .‬ويتأثر االنتشار الكهرومغناطيسي باالستيعاب واالنعكاس من الهياكل واألشياء واألشخاص‪.‬‬‫أال يمكن التنبؤ نظري ًا بالدقة من حيث شدة المجال من أجهزة اإلرسال الثابتة‪ ،‬مثل المحطات القاعدية للهواتف الالسلكية (الخلوية‪ /‬الالسلكية) وأجهزة الراديو المتنقلة البرية‪ ،‬وراديو الهواة‪،‬‬‫والبث اإلذاعي ‪ AM‬و‪ ،FM‬والبث التلفزيوني‪ .‬ولتقييم البيئة الكهرومغناطيسية الناجمة عن مرسالت تردد الراديو الثابتة‪ ،‬ينبغي النظر في استقصاء الموقع الكهرومغناطيسي‪ .‬إذا تجاوزت‬‫شدة المجال المقاسة في الموقع الذي يتم فيه استخدام جهاز التصليب الضوئي مستوى التوافق الالسلكي المطبق أعاله‪ ،‬فيجب مالحظة جهاز التصليب الضوئي فالو كوردليس للتحقق‬‫من عمله بصورة طبيعية‪ .‬إذا لوحظ أداء غير طبيعي‪ ،‬فقد يكون من الضروري اتخاذ تدابير إضافية‪ ،‬مثل إعادة توجيه أو تغيير جهاز التصليب الضوئي‪.‬‬‫بفي نطاق التردد من ‪ 150‬كيلوهيرتز إىل ‪ 80‬ميغاهيرتز‪ ،‬ينبغي أن تقل شدة المجال عن ‪ 3‬فولت‪ /‬متر‪.‬‬‫‪306‬‬‫‪307‬‬

500 hPa

95%

CR123A

10%

CR123A

1060 hPa

LOT

10°C
50°F

40°C
104°F

REF

EN - Importer / DE - Importeur / FR - Importateur / NL - Importeur / IT - Importatore / ES - Importador / PT - Importador / SV - Importör / DA Importør / FI - Maahantuoja / EL - Εισαγωγέας / CS - Dovozce / BG - Вносител / HR - Uvoznik / ET - Importija / HU - Importőr / GA - Allmhaireoir / LV
- Importētājs / LT - Importuotojas / MT - Importatur / RO - Importator / SK - Dovozca / SL - Uvoznik / NO - Importør / PL - Importer / RU - Импортер
/ TR - İthalatçı / AR - ‫ المستورد‬/ TH - ผู้้�นำำ�เข้้า / KO - 수입자 / ZH - 进口商 / JA - 輸入業者

EN - For professional use only / DE - Nur zur Anwendung durch den Zahnarzt / FR - Pour usage professionnel seulement / NL - Alleen voor
professioneel gebruik / IT - Solo per uso professionale / ES - Solo para uso profesional / PT - Apenas para utilização profissional / SV - Endast för
professionell användning / DA - Kun til professionel brug / FI - Vain ammattikäyttöön / EL - Για επαγγελματική χρήση μόνο / CS - Pouze pro odbornépoužití. / BG - Само за професионално ползване / HR - Samo za profesionalnu uporabu / ET - Ainult professionaalseks kasutuseks / HU - Kizárólag
professzionális felhasználásra / GA - Le haghaidh úsáide gairmiúla amháin / LV - Tikai profesionālai lietošanai / LT - Tik profesionaliam naudojimui /MT - Għall-użu professjonali biss / RO - Numai pentru uz profesional / SK - Len na profesionálne použitie / SL - Samo za profesionalno uporabo / NO
- Kun for profesjonell bruk / PL - Wyrób wyłącznie do użytku profesjonalnego. / RU - Только для профессионального использования / TR - Sadece
profesyonel kullanım için / AR - ‫ لالستخدام من قِبل متخصصين فقط‬/ TH - สำำ�หรัับการใช้้งานระดัับมืืออาชีีพเท่่านั้้�น / KO - 전문가 전용 / ZH - 仅供专业人士
使用 / JA - 専門家のみが使用できますMD

EN - Medical Device / DE - Medizinprodukt / FR - Dispositif médical / NL - Medisch instrument / IT - Dispositivo medico / ES - Dispositivo medico/ PT - Dispositivo médico / SV - Medicinska naprava / DA - Medicinsk anordning / FI - Lääketieteellinen laite / EL - Ιατροτεχνολογικό προϊόν /CS - Lékařské zařízen / BG - Медицинско изделие / HR - Medicinski uređaj / ET - Meditsiiniseade / HU - Orvostechnikai eszköz / GA - Uirlis leighis /LV - Medicīniskā ierīce / LT - Medicinos prietaisas / MT - Apparat mediku / RO - Dispozitiv medical / SK - Lekársky prístroj / SL - Medicinska naprava /NO - Medisinsk enhet / PL - Wyrób medyczny / RU - Медицинское Изделие / TR - Tıbbi Cihaz / AR - ‫ جهاز طبي‬/ TH - อุุ ปกรณ์์ทางการแพทย์์์ / KO - 의료기기 / ZH - 医疗器械 / JA - 医療機器

50%

Bigger Lens

CORDLESS BROADBAND L ED CUR I N G L I G HT

For immediate reorder and/or complete descriptions of Ultradent’s product line,refer to Ultradent’s catalog or call toll free 1-800-552-5512.
Outside U.S. call +1-801-572-4200 or visit www.ultradent.com.
Report any serious incident to the manufacturer and the competent authority.

800.552.5512 U L T R A D E N T . C O M
Manufactured by Ultradent Products, Inc. 505 West Ultradent Drive(10200 South), South Jordan, UT 84095.
Made in the USA from globally sourced materials.
Ultradent Products GmbHAm Westhover Berg 30
51149 Cologne Germany
Australian Sponsor:
Ultradent Australia Pty Ltd
22/2 Market St. Sydney, NSW 2000
1800 29 09 29 | www.ultradent.com.au© 2024 Ultradent Products, Inc. All Rights Reserved.
1003883AR11 081224