Advancing the Power Curve®

AC-DC Power Supplies & DC-DC Converters

MedicalProducts for the
Global Medical Environment

MADE IN USA

Medical Grade, Highly Efficient
AC-DC Power Supplies with PFC

P ro du ct Fe at u re s
GENERAL SPECIFICATIONS••High efficiency up to 93% at full rated load current••Delivers up to 1400W of output power (1800W transient)••Semi-regulated output — G-Series & E-Series••Universal 85-264V AC Input Voltage (47-63Hz)••Single output voltages: 12V, 15V, 24V, 28V, 36V, 48V••5V “Always On” standby power output••Active PFC; EN61000-3-2 compliant••Low leakage; EN60601-1 compliant••Low noise; EN55011 / EN55022 Class B compliant••Operating ambient temperature 0-70ºCG-Series

PROTECTION/CONTROL••Over-current, over-voltage, and over-temp protection••DC Power Good and AC Power Good signals••Remote enable input
E-Series

SAFETY••Type B, BF, CF & Defibrillator proof variants available••UL 60601-1:2003••Can/CSA-C22.2 No. 60601-1-08••Can/CSA-C22.2 No. 60601-1-M90••EN 60601-1:2006••EN 60601-1/A2:1995••IEC 60601-1:2005••IEC 60601-1/A2:1995••ANSI/AAMI ES60601-1:2005••EN55011, 55022 and 61000••CE Marked

Output Voltage

G-Series (Single Output)

Power Rating
800W(1000W Transient)

1100W(1300W Transient)

1400W(1800W Transient)

300W(400W Transient)

400W(500W Transient)

500W(700W Transient)(3.00” x 5.00” x 1.46” Open Frame Package)
300W(400W Transient)(3.50” x 5.25” x 1.63” Encased Package)
12V, 24V, 36V or 48V (includes 5V@2A and 12V@4.2A)

400W(500W Transient)

500W(700W Transient)(4.75” x 7.00” x 1.63” Encased Package)
12V, 15V, 24V, 28V, 48V (includes 5V@50mA standby)

E-Series (Single Output)(3.00” x 5.00” x 1.46” Open Frame Package)(3.50” x 5.25” x 1.63” Encased Package)
12V, 24V, 36V or 48V (includes 5V@50mA standby)

E-Series (Triple Output)

E-SeriesCO

N

FIGURATION

SE-Series ACuQor products are available open framed orencased. They can also be double or triple stacked for
additional power output or multiple output voltages*.

E-Series Encased converter
3.50”[88.9 ]

5.25”
[133.4 ]1.63”[41.3 ]

*Accessories including input and output cables are also available for all models and series.

Mix and match a combination of 2 or 3 of any of the E-Series (3” x 5”) packages in either
flat or stacked packages to achieve a higher power and multiple outputs.
Double/Triple Stacked

Double/Triple Flat•• Double Package: S, Thermal Design: C•• Small size: 3.5” x 5.25” x 3.25”•• 600W/800W@ 12V, 24V, 36V or 48V

Double Package: R, Thermal Design: C•• Small size: 6.75” x 5.25” x 1.625”•• 600W/800W/1000W@ 12V, 24V, 36V or 48V•• Triple Package: U, Thermal Design: C•• Small size: 3.5” x 5.25” x 4.875”•• 900W/1200W@ 12V, 24V, 36V or 48V

Triple Package: T, Thermal Design: C•• Small size: 10” x 5.25” x 1.625”•• 900W/1200W/1500W@ 12V, 24V, 36V or 48V

E-Series Open frame converter
3.0”[76.2 ]

5.0”
[127 ]

1.46”
[37 ]

G-SeriesPACKAG

E

G-Series ACuQor products are available
as encased product only. *G-Series converter

7.00”[177.8]6.63”[168.3]4.30”[109.2 ].75”[19.1]

2.28” [57.8]

.16”
[4.1]

4.75” [120.7]

4.84” [123].89” ±.03 [22.5 ±0.7]
INPUT PIN 1

BACK VIEWFRONT VIEW
1.03” ±.04
[26.1 ±1]1.63”[41.3]

Medical Grade DC-DC Converters
Rated for CF Patient Contact and Defibrillation Proof
Pro du ct Fe at u re s
GENERAL SPECIFICATIONS••High Efficiency, up to 93% at full rated load current••Industrial standard quarter-brick pin-out configuration••Low leakage; EN60601-1 compliant

SAFETY••Reinforced Insulation••UL 60601-1:2003••Can/CSA-C22.2 No. 60601-1-08••Can/CSA-C22.2 No. 60601-1-M90••EN 60601-1:2006••EN 60601-1/A2:1995••IEC 60601-1:2005••IEC 60601-1/A2:1995••ANSI/AAMI ES60601-1:2005••EN55011, 55022 and 61000••CF Patient Contact••Defibrillation Proof
Defibrillator proof
Type CF equipment
Output Voltage

CFQorSeries

5V

12V

15V

24V

7A
105W

4A
96W

12Vdc Nominal Input (9-22V Continuous Input Range; 9-25V transient)
Quarter Brick

CF1220A
100W

8A
96W

24Vdc Nominal Input (18-36V Continuous Input Range; 18-50V transient)
Quarter Brick

CF2424A120W

10A120W

8A120W5A
120W

48Vdc Nominal Input (34-75V Continuous Input Range; 34-100V transient)
Quarter Brick

CF4825A
125W

12A
144W

10A
150W

6A
144W

PACKAGESThe CFQor series of Quarter-Brick DC-DC converters
is designed for medical applications that require
isolation and leakage current levels that comply
with IEC60601-1 for CF patient contact & are also
defibrillation proof.
They are high efficiency, fixed switching frequency
converters based on SynQor’s well proven and reliableindustrial grade products. CFQor products are fully
encased for additional environmental protection.

1.54”(39.01)

FlangedQuarter Brick

Quarter Brick

2.39”(60.6)

Defibrillator proof
Type CF equipment

Converterheight0.5”(12.7)

Advancing The Power Curve®
Founded in 1997, SynQor ® quickly became the technology, quality and service
leader for high efficiency dc-dc converters for the telecom/datacom marketplace.
The PowerQor ®, BusQor ®, DualQor ®, iQor ™, and NiQor ® product lines, combinedwith SynQor’s unmatched lead-times, flexibility, design support and lowest total
cost of ownership are the benchmark sought by SynQor’s worldwide customers.
In 2004, SynQor began its successful entry into the industrial, railway, medical
and military market segments with the InQor®, RailQor®, ACuQor®, MilQor® andUPS product lines. These new technology leading dc-dc and ac-dc productlines have enabled customers to realize vast savings in space, weight and overallsystem costs while dramatically increasing power densities and end-system
performance. Equally important is the service and support advantage SynQor
has brought to customers in these market segments.
SynQor is a privately owned U.S. company headquartered in Boxborough,Massachusetts, which is also the location of all of its manufacturing operations.
SynQor has a design center in Dallas, Texas, and sales/marketing offices
throughout the USA, Europe, Malaysia, Korea and China. More information isavailable on the company’s website at www.synqor.com.

SynQor Headquarters
155 Swanson Road Boxborough, MA 01719-1316
Phone: 978-849-0600 Fax: 978-849-0602
Toll Free (USA): 888-567-9596
E-mail: mqnbofae@synqor.com
www.SynQor.com

006-0005958 - 12/01/2015 - Rev C

Made in USA