DIGITAL CALIPERS OFFSET TYPEINSTRUCTION MANUAL
Model No. : D-150F

Thank you for purchasing the Digital Calipers Offset Type.
Before use, please read this document thoroughly for proper operation.

PART IDENTIFICATION
123 459

1. Jaw
2. LCD Display
3. Lock Screw
4. [Hold] Button
5. Connector (*)
6. [ON / OFF] Button

7. [ZERO] Button
8. Battery Cover
9. Scale
10. Shaft
11. Jaw Lock Screw(*) Output connector function not available for this model.

11167810NOTES・Before use, please wipe off corrosion protectant using a soft, oiled cloth.
・This is a precision instrument, handle with care. Do not drop or subject to excessive force.
・Do not damage scale or scratch instrument. Do not apply ID numbers with engraver.
・Keep away from direct sunlight. Do not store instrument in extreme temperatures.
・Do not use in locations with high electric fields, such as near fluorescent lights or switching power supplies -- may cause erroneous
reading to display on LCD.
・If instrument will not be used for a period of one month or more, please remove the battery to prevent damage from battery fluid leakage.

TROUBLESHOOTINGBATTERY REPLACEMENT
1. Slide cover in direction of arrow.
2. Insert battery with (＋) sideoutward.
3. Replace battery cover.
CAUTIONBatteryCoverOpen

Only use SR44 type battery.

Q. Nothing is displayed, or the numbers are flashing.
A. The battery may be running low, please replace battery
with a new one.

Q. With power on, the number on the LCD does notchange as the carriage is moved.
A. Please reset by removing battery for about 30sec., andthen replacing.

Q. Nothing happens when buttons are pressed.
A. Make sure battery is properly inserted.

OPERATION
Absolute Measurement

1. Turn on power.
2. Press [ZERO] button with caliper jaws fully closed.
3. When display reads “0.00mm”, caliper is ready for use.

Relative Measurement

1. Turn on power.
2. Open the caliper jaws to desired reference value and press[ZERO] Button (OK at any position.)
3. When display reads “0.00mm”, caliper is ready for use.
CAUTIONIf [HOLD] Button is pressed for a long period, LCD may display
characters such as “MIN”, “MAX”, and “△”. To clear display, please cyclepower, or press and hold the [HOLD] Button several times until clear.

EXAMPLE OF USE
Loosen the Jaw Lock Screw and adjust position of the jaw as
required by step in the object to be measured. Tighten the Jaw
Lock Screw before taking a measurement.

Q. The characters “MIN”, “MAX”, or “△” are displayed
on LCD.
A. Cycle power, or apply several long presses to [HOLD]
Button. Use as normal once character is off.

SPECIFICATIONS・Measuring Range : 0−150mm
・Resolution：0.01mm
・Instrument Error : ±0.04mm
・Maximum Response Speed : Moving Slide to Right…1.5m/s
Moving Slide to Lef t …0.75m/s※“Right” and “Left” when viewed as shown in PARTS IDENTIFICATION
drawing.

・Operating Temperature Range : 0℃〜40℃
・Storage Temperature Range : −10℃〜60℃
・Auto Off : 15〜25 minutes
・Power : Silver Oxide Battery, SR44, 1.55V・Battery Life : Approx. 1 year with normal usage
・Accessories : Silver Oxide Battery, SR44, 1.55V, (for test)

Niigata seiki Co., Ltd.
6-15-22, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-31-5660 Fax. : +81-256-39-7730

MAIL:intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

I258-K 1510