Thank you for purchasing the Digital 3-Point Micrometer.
Please read this manual thoroughly before use for proper operation.

FEATURES● Easy to read digital display.

● High accuracy 3-point ID measurements.

● Includes extension rod (Deep Hole Extension Rod) for measurements in deep measurements.

Gauge Tips

● Probe Material： Carbide tipped● Auto-display-off function (approx. 5 minutes.)…SET

MCD3385-0608HTN
MCD3385-0810HTN
MCD3385-1012HTN
MCD3385-1216HTN
MCD3385-1620HTN
MCD3385-2025HTN
MCD3385-2530HTN
MCD3385-3040HTNMCD3385-4050HTNMCD3385-5063HTNMCD3385-6275HTN
MCD3385-7588HTN
MCD3385-87100HTN

6− 88 − 1010 − 12
12− 1616− 2020− 2525− 3030− 4040− 5050− 6362− 7575− 8887−100180

Use the following procedure and the special spanners provided to attach the deep holeextension rod.

[180[ABS/INC…UNIT] Button

230

4…SET] Button

1. Place the accessory spanners into the notches above and below the split in the rod
and turn to separate the rod into two parts, the “gauge head” and “display unit.”
See Figure 2
2. Attach the extension rod to the measurement head and then attach to the display
unit. Tighten with accessory spanners. See Figure 3
3. Re-set the origin using the procedure above -- “Setting the origin”

Battery Cover 【BACK】

230300320420510

Dedicated Spanner (2x)Attach5Deep Hole Extension Rod…SET720

Gauge Head

SR449901010

SR44 (Silver Oxide Battery)Less than 285

CDCB

A100

105
Less than 1 110

Deep Hole Extension RodConfirm displayed
value matches
ring gauge dimensions.

1Figure 2RodNotchLoosenSpanner

ABS/INC…UNIT

Display Unit

1. Remove any grease and dust from the measuring surfaces with a clean cloth, such as a lens cleaning cloth. Make sure no foreign objects
are adhered to the surfaces.
2. Press the [ …SET ] button to turn on power.
3. Make sure “ABS” is displayed on the LCD; press the [ABS/INC…UNIT] if required.
4. Insert the measurement head into object to be measured and turn ratchet thimble until the measuring surfaces make contact with the inside
diameter. Continue turning until ratchet “clicks” 1~3 times and then read the measurement off the LCD display.

TROUBLESHOOTING43

ERROR CONDITION

150
B

CORRECTIVE ACTION

Measured value is wrong● Check if measuring surfaces are dirty. Clean with lens cloth.

Gauge head does not retract● When stored at low temperature, or not used for a while, the internal grease may become stiff.

● Check to insure origin-point is correct. → refer to section “Setting the origin”

Allow sufficient time for grease to warm up.

[ Deep Hole Extension Rod ]● Wearing dust-free gloves, use hand to push in.

● Nothing is displayed on LCD● Display is not stableINSERTING THE BATTERYOpen

3 Press

Press the [ABS/INC…UNIT] button to reset display to zero while retaining the origin setting in memory.
To return to normal (absolute) measurement mode, press the [ABS/INC…UNIT] button again.

[ Body : MCD3385-5063HTN〜MCD3385-87100HTN ]A

127

ABS/INC…UNIT

INC Mode

B

100…SETOPERATION※Battery provided for testing.
Please replace with fresh

Battery Cover Tool

[ Body : MCD3385-0608HTN〜MCD3385-4050HTN ]A

Ratchet Thimble

Figure 3Attach710

DBattery Cover2 TurnFigure 1

DEEP HOLE EXTENSION ROD

Rod

180

Dimensions (mm)Model No.
MCD3385-0608HTN
MCD3385-0810HTN
MCD3385-1012HTN
MCD3385-1216HTN
MCD3385-1620HTN
MCD3385-2025HTN
MCD3385-2530HTN
MCD3385-3040HTNMCD3385-4050HTNMCD3385-5063HTNMCD3385-6275HTN
MCD3385-7588HTN
MCD3385-87100HTN

Gauge Head

ABS/INC…UNIT

C

Model No.

Maximum Weight
Error (µm)(g)

1. Prepare instrument and ring gauge by cleaning measurement head and ring gauge
inner diameter with appropriate cleaner.
2. Press the [ …SET ] button to turn on power.
3. Press and hold the [ …SET ] button for more than thee seconds, “Set” will display
on LCD and will start blinking when you release the button.
Confirm that the displayed value matches the dimensions of the ring gauge standard.
If they do not match, use the following procedure to modify.

4. Insert the measurement head into the ring gauge and turn the thimble until themeasuring surfaces make contact with the inside of the ring gauge. Continue turning
until ratchet stop “clicks” 1~3 times. See Figure 1
5. Press the [ …SET ] button. The flashing “Set” will disappear. The displayed value
should match the dimension of the ring gauge standard, and origin setting iscomplete.

● Power：SR44（Silver Oxide Battery）※1x included for testing

Measuring
Range (mm)“Click, Click”

InsertRatchet Thimble(constant pressure mechanism)
LCD Display

PART IDENTIFICATION● Resolution：0.001mm

Set the origin to be used for measurement.
The origin is stored even when power is off, so it does not have to be reset every time
power is turned on. When the battery is replaced, the origin will have to be reset.

With “Set” flashing on LCD, a long press on the [ …SET ] button will move theblinking to the readout digits in sequence. A short press on the [ …SET ] button willchange the value at the cursor. Repeat until correct value has been entered and “Set”
blinks.

● With constant pressure ratchet mechanism.

SPECIFICATIONS※Prepare a Ring Gauge to use as a reference standard

ABS/INC…UNIT

Part Number : MCD3385-**HTN<combined>

SETTING THE ORIGIN…SETINSTRUCTION MANUALELECTRONIC DIGITAL THREE-POINT
INTERNAL MICROMETERS

1. Using the battery cover tool, open the battery cover by rotating in the directionof the arrow.
2. Insert battery with “＋” side outward.
3. Replace battery cover.

PLEASE NOTE
Use only SR44 type battery.

NOTICE
Micrometer is a precision instrument. For specified performance, use in environment around 20°
C and allow sufficient time to acclimatize.
Temperature extremes may affect instrument accuracy and operation.

● Display is not clear● Remove and reinsert battery to cycle power.
● Replace the battery.

※If error continues, or if you have any questions, please contact distributor or place of purchase.
※Please note, manufacturer is unable to respond to inquires or provide service directly. Please contact distributor or place of purchase.

CAUTION● Always remove dirt from measuring surfaces with a lens cloth before use.
● Do not turn thimble more than 3mm above upper measurement range.
● This is a precision instrument, handle with care. Do not drop or subject to excessive forces.
● Keep instrument away from direct sunlight or high temperatures such as in a car, or near a stove or heat source.
● Keep instrument away from water and oil.
● Please do not disassemble or modify.
● Remove dust and cutting chips after use and apply rust preventative oil to metallic surfaces. Please keep out ofreach of children.

● When instrument will be idle for one month or more, please remove the battery to prevent damage from battery

fluid leakage.
● Please do not use organic solvents to clean the instrument body.
● Only use as directed. Improper use may cause accident or injury.

Niigata seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

品番：MCD3385−＊＊HTN<兼用>

この度はデジタル三点マイクロメータシリーズをお買上げいただきありがとうございます。
ご使用前に、本書をよくお読みいただき正しくお使いくださいますようお願い申し上げます。

②“● 継ぎ足して使用できる「深穴測定用延長ロッド」付属で、より深い箇所の測定も可能です。
● 定圧装置付● 測定子材質：超硬ラチェットシンブル● 電源：SR44（酸化銀電池）※テスト用１個付属● オートディスプレイオフ機能付（約5分）

品

番…SET

測定範囲 最大許容 質量（mm） 誤差（μm） （g）

ロッド

180
180

4ABS/INC…UNITボタン

230230電池蓋【裏面】

300320420510

専用スパナ（２個）

深穴測定用延長ロッド

710

5990
1010

電池蓋開閉スパナ

深穴測定用延長ロッドを使用する場合は、必ず専用スパナを２個使用し、下記の手順で取り付けてください。

番Ａ

取り付ける

※付属の電池はテスト用です。ご使用の際は新たにお買い求めください。

ＣＤ
C

MCD3385-0608HTN
MCD3385-0810HTN

2以下A

100

85

MCD3385-1012HTN

ゆるめる取り付ける100

MCD3385-2530HTN

43

1以下 110150

MCD3385-5063HTNMCD3385-6275HTN
MCD3385-7588HTN

A

127[ 深穴測定用延長ロッド]

MCD3385-87100HTN

開ける

電池は必ずSR44をお使いください。

マイクロメータは非常に緻密に作られており、その性能を発揮するためには20℃前後に温度慣らしをした状態でご使用ください。
極端な温度下では測定値の精度や測定器の動作に影響が出る場合がございます。

ＬＣＤ部側

図２

図３

…SET”ボタンを押して、電源を入れてください。

③“ABS/INC…UNIT”ボタンを押し、「ABS」を表示させてください。

“ABS/INC…UNIT”ボタンを押すことで、原点を維持したまま表示をゼロリセットすることができます。
原点からの測定値に戻りたい場合は、再度“ABS/INC…UNIT”ボタンを押してください。
トラブルの内容

対応策

測定値が間違っている● 測定面の汚れをグラスペーパなどで拭き取ってください。

測定子が戻らない● 低温で使用したり、しばらく使用していないと、内部のグリスが馴染んでいない可能性があります。十分温度慣● 原点が合っているか確認してください。→『原点の設定』参照

らしを行ってください。

● 手袋を着用し、手で押し込んでください。
● 電池を入れ直してください。

● 新しい電池と交換してください。

※症状が改善されない場合や不明な点がございましたらお買い上げの販売店、または発売元までご連絡ください。
※お問い合わせや、ご連絡が無いまま直接発売元に修理品などを送付されても処理、対応ができない場合がありますのでご了承ください。

ご注意

使用上のご注意①本体裏面の電池蓋を、矢印の向きに付属の電池蓋開閉スパナなどで回して開けます。
②電池を、必ず（＋）側を表にしてセットしてください。
③電池蓋を元通りに装着してください。
ご注意②“● LCD表示部に何も表示されない● 表示が安定しない● 表示が不明確である

D

電池の入れ方

電池蓋

B

トラブルと対応

MCD3385-4050HTN[ 本体：MCD3385-5063HTN 〜 MCD3385-87100HTN ]C

105

ABS/INC…UNIT①測定子を含めた測定部を脱脂し、グラスペーパなどで拭き取って異物が付着していない状態にしてください。

INC

MCD3385-2025HTN

モード

MCD3385-1620HTN

深穴測定用延長ロッド

スパナ④測定部を被測定物の内側に入れ、ラチェットシンブルを持って回します。測定子と被測定物の内径が接触し、ラチェットシンブルが『カチカチ』と鳴ったところから１〜３回転させたところで停止して、LCD表示部の数値を読み取ってください。

B

MCD3385-1216HTN

MCD3385-3040HTN

測定部側[ 本体：MCD3385-0608HTN 〜 MCD3385-4050HTN ]Ｂ

切欠②ロッドの「測定部側」と「LCD部側」の間に深穴測定用延長ロッドを取り付け、専用スパナで締めてください。 図３SR44（酸化銀電池）

使用方法

品

ロッド

①付属の専用スパナをロッドの切欠に差し込んで回し、ロッドを「測定部側」と「LCD部側」に分離させてください。 図２…SET

寸法図（単位：ｍｍ）

図１③上記『原点の設定』に従って、原点を設定してください。

ＳＲ４４

720

ABS/INC…UNIT

リングゲージ寸法値と数値が一致しているか確認

１⑤“ …SET ”ボタンを押してください。
LCD表示部の「Set」の点滅が消え、リングゲージに表示されている寸法値とLCD表示部の数字が一致していれば、原点設定の完了です。

…SET ボタン

180…SET

３ 押す④測定部をリングゲージに入れ、ラチェットシンブルを持って回します。
測定子とリングゲージの内径が接触し、ラチェットシンブルが『カチカチ』と鳴ったところから１〜３回転させたところで停止してください。 図１

ABS/INC…UNIT

深穴測定用延長ロッド

6- 8
8- 10
MCD3385-0810HTN
10- 12
MCD3385-1012HTN
12- 16
MCD3385-1216HTN
16- 20MCD3385-1620HTN
20- 25
MCD3385-2025HTN
MCD3385-2530HTN 25- 30
MCD3385-3040HTN 30- 40
MCD3385-4050HTN 40- 50
50- 63
MCD3385-5063HTN
62- 75
MCD3385-6275HTN
MCD3385-7588HTN 75- 88
MCD3385-87100HTN 87- 100
MCD3385-0608HTN

測定部”ボタンを押して、電源を入れてください。

「Set」の文字が点滅している状態で、“ …SET ”ボタンを長押しすると、点滅位置が変わります。
点滅している位置で、一度“ …SET ”ボタンを離し、再度短く押すと数字が変わります。
これを繰り返し、リングゲージに表示されている寸法値に数値を設定したら、点滅位置を「Set」に戻してください。

（定圧装置）

LCD表示部

測定子

…SET

入れる

● 最小読取値：0.001mm

２ 回す③“ …SET ”ボタンを3秒以上長押しし、離すとLCD表示部の「Set」の文字が点滅します。
ここで、LCD表示部の数字とリングゲージに表示されている寸法値とが一致していることを確認してください。
一致していない場合は、下記の手順で合わせてください。

● 内径を３点で測定するため、より精度の高い測定が可能です。

各部の名称

ラチェットシンブル

ABS/INC…UNIT

長様

カチカチ

①測定子を含めた測定部と、基準リングゲージの内径をパーツクリーナーなどの洗浄剤で洗浄します。

● デジタル表示で見やすく、読み取り間違いがありません。

仕※測定前に基準リングゲージをご用意ください。

通常測定で使用する原点を設定します。
この原点は電源をOFFにしても記憶されているので、毎回設定し直す必要はありません。
電池交換後はリセットされますので、再度設定し直してください。

…SET

特原点の設定デジタル三点マイクロメータ

取扱説明書● ご使用の前に、必ずグラスペーパなどで測定子の汚れを除去してください。
● 測定範囲の上限を３mm以上上回ってスピンドルを回さないでください。
● 精密機器ですので、落としたりぶつけたり、強い力を加えたりしないでください。
● 直射日光の当たる場所や炎天下の車の中、火のそばなどでの使用、保管はしないでください。
● 水や油のかかるところで使用しないでください。
● 分解・改造をしないでください。
● 使用後は柔らかい布などでゴミや切粉などを取り除き、防錆油を塗布してお子様の手の届かない場所へ保管してください。
● １ヵ月以上ご使用になられない場合は、本器より電池を取り外して保管してください。電池の液漏れで本器を破損する恐れがあります。

● 本体の手入れに有機溶剤を使用しないでください。
● 用途以外のご使用は、事故やけがの原因となりますので絶対におやめください。

発売元〒955-0055 新潟県三条市塚野目5丁目3番14号(0256) 33-5502(代) FAX (0256) 33-5528

URL http://www.niigataseiki.co.jp

E364-T

20111000