For information about softwre support and version update, please visit our web site

http://www.niigataseiki.net/official/english/support/software_slog.html

User ManualModelNo.

DTC-1/DTC-2

Android Version Download

S-LINE DATA TRANSFER CABLE & SOFTWARE※This software is also available for Android.
Please use the QR code or URL shown.

※Only for models that support USB host mode capability.
Host USB cable required.

Thank you for purchasing S-Line Data Transfer Cable. Using the supplied software ( “SLOG”,) measurement data can be sent to the PC for capture and storage.
Please use the following procedure to install the software on your PC .

https://play.google.com/store/apps/details?id=jp.co.niigataseiki.slog

DRIVER INSTALLATION ※ Installation requires Administrator privileges for the computer.
the supplied disk into the CD-ROM drive of your computer.
1 ①Insert②Using the data transfer cable (DTC-1, DTC-2), connect the device to the computer. Note: Make sure the USB connector is oriented properly, and insert as far as it will go.
③The “Found New Hardware Wizard” screen will display as shown below.

2 Click on "Close"

Right-click on the named device located under
3 "USB
Serial Port" in the device manager.

4 Select"Browse my computer for driver software".

"Include subfolders" and click on
5 Check[Browse] button,and select "Update Driver Software...".

then select the CD-ROM Drive from the disk
provided and click on "OK." With driver location
selected, click on "Next."

CD-ROM Drive SelectClose SelectInclude subfolders Check

Update Driver Software...
SelectBrowse my computer for driver software
Selectinstallation is complete, click on
6 When"Close" to close the wizard.

Installation is complete.

7 Depending on USB configuration of the computer, there are times whenFound New Hardware screen will display a second time.
In this case, repeat the procedure above starting with section 3 .
Please note, If you do not install drivers a second time if requested,
the USB port recognition may be unstable.

Next SelectSpecifications
Cable Length

Approx. 3mConnector

USB

Compatible OS Windows XP ・ Vista ・ 7 ・ 8.1 ・ 10
Compatible Models

DTC-1

8To make sure a virtual COM port has been implemented,
please open the Device Manager and click on the "+" to the left of "Ports(COM & LPT)" and confirm.

DIGITAL S-LINE CALIPERS D-***IP67S
ELECTRONIC DIGITAL S-LINE MICROMETERS MCD-***IP67S
DIGITAL S-LINE INDICATOR DEI-WKS*

DTC-2

COM port has been added.

※COM port number will vary depending on PC.

To capture and record measurement data, please go to the procedure on the

9 reverse side -- “Using the data transfer software (SLOG.)”
Close Select※Disconnect the data transfer cable from the PC USB port, and proceed tosoftware configuration steps on the reverse side.

DIGITAL S-LINE INDICATOR DEI-233S*
DIGITAL S-LINE TEST INDICATOR DET-234S*
ELECTRONIC DIGITAL S-LINE THREE-POINT INTERNAL MICROMETERS MCD-****IPS

Niigata seiki Co., Ltd.
6-15-22, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-31-5660 Fax. : +81-256-39-7730

MAIL:intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

I249-K11512

USING “SLOG” DATA TRANSFER SOFTWAREMISCELLANEOUS SETTINGS is disconnected (when measurement window is closed.)

1 Preparations Before Use

5 FilePlease only change these setting when communication●Modify Color①Please install the drivers on your computer following the “Driver Installation” procedure on the reverse side.
②Copy “SLOG.exe” from the CD-R disk to your computer and launch the software by double-clicking.
③Connect the data transfer cable to the measuring instrument data output port.
④Connect the data transfer cable to the PC USB port.

You can change the colors of the following:
・ SLOG display window, background andcharacters.
・ In digital display measurement screen,display bar color for inside, and outside
tolerance range.

Connect data transfer cable in the following order: 1st -- measuring instrument, 2nd -- PC.

2Setting the Output Destination for the Data. (When recording measurement to other applications.)①Please launch the application that you want to transfer (record) the data to.

●Language(Internet Explorer 8 or greater is required.)②Place a check next to “Output Window” and click on the [Regist...] button.
Running applications will be displayed in a list.
Select the desired application and click on the [OK] button.

5293

English and Japanese options available.

6 HelpThis User Manual is also available via browser.

Destination application: Support display of text.
Examples: text editor (Notepad,...) , Microsoft Excel, Microsoft Word, etc.

671087 Data Received●BeepWhen checked, “beep” will sound when datais recorded.

Select●Blink411

When checked, the screen will flash when data is recorded.

8 Key to be added

3 Selecting Receive TypeIf checked, a key control character can be automatically sent after recording data.
The key can be selected from the following options: [ENTER], [TAB], [DOWN] ,[UP], [LEFT], or [RIGHT]

Please chose the appropriate signal reception method corresponding to the measuring instrument.
There are “Single Receive” and “Real Receive” options available.

● Single Receive

Applicable Instruments：DIGITAL S-LINE CALIPERS D-***IP67S
ELECTRONIC DIGITAL S-LINE MICROMETERS MCD-***
Measurement Screen：Fixed number display(Measurement value changes only at time of recording.)

How to Record ：Press the [SET] button on the measuring instrument.

● Real ReceiveData
4 Start Receiving[]

Applicable Instruments： All instruments compatible with DTC-1 / DTC-2
Data Transfer Cables.
Measurement Screen： Real time display
How to Record ： Click on [Record] on SLOG measurement screen,or press [F9] Key

Example -- With output destination set to Excel, if you select the key Type [DOWN]⇒ After recording data the cursor will automatically move down one cell on the spreadsheet.

9 Display●Unit

Measured value rounded to the specified significant digits before display and recording: 0.1, 0.01, 0.001, 0.0001●Kind

You can select between two types of display for the measurement screen: Digital Display or Analog Display.

Digital Display

Analog Display

Click on the Start button on SLOG. Measurement screen will open and begin displaying the data from the measuring instrument.
Record data using the method listed under “How to Record” in section 3 above.
※ If destination application is minimized, data will not be recorded. Please open the destination application to record data.
※ Click on the [Stop] button or close the measurement window to disconnect communication. To resume, click on the [Start] button
again. To close the SLOG software, click on the [End] button.

10 Tolerance※Measuring instrument of power OFF, connecting or disconnecting the cable, when replacing the batteries, please stop the communication of once SLOG.

When checked, the display will change color to indicate
if a measurement is within the tolerance range set below.

◎When more than one measuring instrument is connected to the PC●Nominal …………… Enter the reference value for the dimension tolerance.

●Lower deviation … Enter the lower value for the dimension measurement.

If you are connecting multiple instrument to the computer, a list for
selecting a connection destination is displayed.
Select the instrument you want to connect, please press the[Connect] button.
※ Some of the measuring instrument there is equipment there is noability to identify the equipment, you might be displayed as "undefined".
※ You may not appear in the list by state of the connection timing andmeasuring instrument cable. In that case, once you press the [Cancel]
button, please press the [Start] button on the main screen again.

●Upper deviation … Enter the upper value for the dimension measurement.

※ Please set the tolerance numbers such that Upper Tolerance > Lower Tolerance.
※ Allowable tolerance range is from -1000 〜 1000
※ The color of the display can be set in section 6 above, "Modify Color,"
to indicate if a measured value is within tolerance range by changing the color of the displayed bars.

Select the device you want connect

Selectto hide the SLOG window.
11 Hide Click
To see the window again, click on the SLOG iconin the windows task bar.