PROFESSIONAL

INSTRUCTION MANUAL
Model No. WDI-3058SP WDI-8058SP
WDI-5058SP WDI-10078SP

SAFETY PRECAUTIONS

Please Observe

Always follow the procedures specified below in order to prevent harm to yourself or others, and toprevent damage to property.

LONG STROKE DIAL GAUGE■Content marked as follows indicates risk of injury or damage if not followed.

Thank you for purchasing the Niigata Seiki Long Stroke Dial Gauge.
Used with a Magnetic Base or Indicator Stand, this gauge will show the difference in heightor position relative to a zero point set at a reference position.

■These symbols mark content that must be observed.

■APPLICATIONS・Comparing to a master part during inspection.
・Measuring machine tool positioning accuracy.
・Measuring runout for rotary shafts.

・Checking vise parallelism on milling machines.
・Measuring flatness of surfaces and assemblies.
・Confirming machine tool feed distance.

●For safe and proper use of this product, please read this
instruction manual before use and follow the procedures
described. Please keep manual where it is accessible touser for future reference.
●Keep this manual with the instrument if transferred orleased to a third party.
●For inquiries about this product, please contact dealer orNiigata Seiki at the address listed on the following page.

■PARTS IDENTIFICATION & FUNCTION① LimitMarkers② Pointer(Needle)In this manual,
indicates RISK OFPERSONAL INJURY OR PROPERTYDAMAGE if not followed. The
symbolindicates something which is PROHIBITED,and thesymbol Indicates REQUIRED
step or necessary condition.
※Model WDI-3058SP shown⑩ Flat Rear plate⑨ Bezel090

SAFETY NOTIFICATIONS

10
0

80

25

WDI-3058SP[Accessories]

20

5

70

20

0-30mmJEWELEDSHOCK-PROOF

10

30

15

60

40
50

③ Main Scale⑦ Spindle⑧ Contact Point(Probe)⑪ Lug④ Revolution
Counter

0.01mm

⑫ Mounting Hole(6.5mm）⑤ Rev. CounterScale⑬ Rear Plate
With Lug⑥ Stem※WDI-3058SP・WDI-5058SP・
WDI-8058SP……… 8ｍｍ※WDI-10078SP ……10mm①Limit Markers ········ For marking acceptance
range for Pass/Fail testing.
②Pointer (Needle) ··· Shows measurement on③Main Scale ············ Main Scale. rotate Bezel (⑨)
to turn.
④Revolution Counter ·· Pointer indicates Main Scale
x100⑤Rev. Counter Scale ·· Each division is Main Scalex100⑥Stem ····················· For holding and mounting
Gauge.

※To Use the Rear Plate with Lug,
unscrew the standard Flat Rear
Plate screws, replace, and use
screws to secure.

⑦Spindle ····················· Shaft moves up and downwith measurement.
⑧Contact Point ··········· Probe which contacts
workpiece.
⑨Bezel ························ Rotate to turn Main Scale(③)⑩Flat Rear Plate ········· Standard back cover.
⑪Lug ··························· Alternate mounting point.
⑫Mounting Hole ········· For mounting Gauge.
⑬Rear Plate with Lug ·· Accessory back cover for
mounting Gauge using Lug.

HOW TO USE - Preparation

Mounting Gauge by other than Stem or Lug will cause inaccuracy and product damage.

3. Confirm that Pointer and Revolution Counter movement is smooth.
Using fingertip, gently press on Contact Point to move the Spindle up and down. Motion of Pointersshould be smooth. If it is not smooth make sure Stem is not clamped too tight, and adjust. Also makesure Pointer is stable at set position.

4. Make sure Spindle axis is perpendicular tomeasured surface.

Read the manual and follow all instructions.

Do not disassemble or modify.

・Use of product other than as described in the
manual may cause accident.

・It may damage Gauge and cause
poor accuracy

Use only as Indicator Gauge.

・Use for any purpose other than measuring may
damage or wear the instrument. Improper use
may also cause accident.

Do not shock Spindle.

・Rapid motion, or lateral force may
damage Gauge and cause poor
accuracy.

Use in an environment which meets thefollowing conditions:●Temperature within range of 0~40°
C, humidity
30~70% (non-condensing.)●Location with minimal dust, oil, oil mist, andprotected from direct sunlight.
●Location protected from use by children andunauthorized people.
・Use in location contrary to the above may cause
poor accuracy, damage to the product, or may
result in accident or injury.

No Rapid
MotionHandle With Care.

・Do not drop or subject to shock, do not place
under heavy objects. Damage may cause failure
or poor accuracy.

No Lateral
Force

PREPARATION - Mounting
Dial gauge must be securely mounted such as on a comparator stand or magnetic base.
Please follow these guidelines.

Gauge Holder

Make sure Gauge Holder is rigid.

・Holder must be sufficiently secure to prevent deflection fromthe weight of the Gauge.
・Holder must be rigid enough to hold Gauge and not to liftfrom measurement force.
・Holder support arm should be as short as possible to prevent
deflection.
※Deflection or lifting will cause measurement error such as
origin position error and inaccuracies in measured reading.









:',63










PP

⑥ StemPP-(:(/('
6+2&.3522)







SupportArm

Minimize

Dial Gauge must only be attached by Stem or Rear Lug.

・Mounting of gauge by other than Stem or Lug will cause
inaccuracy and product damage.

During installation, do not over-tighten the Stem.

・Excessive force on the Stem may cause Spindle to bind.

※Mounting Example■Measurement is not stable, or measurement accuracy is poor.

・Contact Point may be worn.
Worn Contact Point will affect accuracy. Periodically check for wear, and replace if wear is affecting
measurement accuracy.

SPECIFICATIONS
Right
angle

Model No.
Measured Object

Measured Object

HOW TO USE - Comparison Measurements

1/10 Rev Indication Error Repeat- Weight
Gradu- Measurement
Hysteresis Measuring
1
the Entire
Indication overMeasuringability (For Flat
Scale Error
Range
ation
Force Revolution Error
Range
(mm)

(mm)

(μm)

WDI-3058SP
WDI-5058SP
0.01
WDI-8058SP
WDI-10078SP

30
50
80
100

7
9
11
15

0 -100●Replacement Contact Point Part No.: DI-CP

(N)

≦ 3.0
≦ 3.5
≦ 4.0
≦ 4.5

(mm)

(μm)

1

14
15
―
―

(μm)

(μm)

35
40

5

50

6

Rear Plate)

210g
270g
290g
530g

Ex.) Pass Range:- 0.05 to +0.15mm① Limit Markers

AFTER USE CARE, STORAGE
Remove any dust or dirt after use. ※ Do not lubricate.

2. Set the Origin.
Adjust the gauge mount or rotate the Bezel to set the
Gauge to "0"0025② Pointer※Setting the Limit Markers

80

20
5

WDI-3058SP

30

0.01mm20

・Wipe any contamination from Spindle sliding surface using a dry cloth, or cloth moistened withalcohol.
・To clean other surfaces, wipe with a soft dry cloth, or a cloth moistened with a mild cleaner.

1090

3. Remove Reference, and begin measurements.
Remove Reference or Master, careful not toshock Spindle. Insert part to be measured andread the measurement off the Scale.

Caution・Temperature changes during measurement can cause repeatability error.
Please try the following solutions:●Use in location with constant temperature.
●When taking measurements, periodically adjust zero point using a Master reference to correct for
temperature induced drift.

2. Attach to the Gauge Holder using the Stem or Rear Lug.

Carefully lift Spindle with fingertip, and, taking care
not to hit Spindle from the side, insert the ReferencePart or Master under Contact Point.

Denotes a prohibition – You MUST NOT do■Origin position shifts during measurement.

If loose, tighten Contact Point and Rear Plate Screws.

1. Set Up Reference Part.

Denotes a requirement – You MUST do

TROUBLESHOOTING

1. Confirm that the Contact Point and Rear Cover are tightly fastened.

If Spindle (Contact Point) is not at a right angle tosurface, Gauge will not operate properly andmeasurement will be inaccurate. Always keep theSpindle axis perpendicular.
※When used to check parallelism of Milling Machinevise, use a Magnetic Base to mount the Gauge
perpendicular to surface, and move it out of the way
during operation to prevent interference.

WARNING Indicates risk of personal injury or property damage if not followed.

0-30mmJEWELEDSHOCK-PROOF

70

10

40

15

60

50

Limit Markers can be moved to show acceptance
range for measurements.

HOW TO USE - Parallelism, Flatness, Runout, etc.
1. Position Contact Point on surface.
Carefully lifting Spindle with fingertip, and taking care not to hit Spindle from the side, position
the surface to be measured under the Contact Point.

Check for wear of Contact Point.

・Measurement accuracy will be affected by worn Contact Point. Regularly check for wear andreplace Contact Point if worn.

Store in provided case in a cool, dark, and dry location.

・During storage, make sure there is no force on the Spindle (such as pushed in, or lateral force.)
・Keep away from moisture and direct sunlight, and secure from unauthorized personnel.

CALIBRATION●In order to maintain instrument accuracy, it is recommended
that accuracy is confirmed through calibration on a periodic
basis.
Wear of measuring surfaces from repeated use may affectaccuracy and periodic accuracy checks should be performed.

2. Set the Origin.
Adjust the gauge mount, or rotate the Bezel to set the Gauge to "0"

3. Read the scale as the measured part is moved.
Slowly move the part while monitoring the Pointer and reading the measurement.

We provide calibration services. Please contact
agent in country of purchase to make arrangements.

Niigata Seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

業務用

取扱説明書

WDI-3058SP WDI-8058SP
WDI-5058SP WDI-10078SP

品番

ロングストロークダイヤルゲージ

お使いになる人や他の人への危害、財産への損害を未然に防止するため、必ずお守りいただくことを下記のように説明しています。

注 意 『傷害を負う、または物的損害が発生するおそれがある内容』です。

■お守りいただく内容の種類を次の図記号で区分しています。
必ず実行していただく内容『強制事項』です。

・フライス盤のバイス平行確認・加工、組立における面の測定
・機械加工における送り量の確認

・工作機械の精度検査
・回転軸の振れ検査●正しく安全にお使いいただくため、ご使用の前に本取扱説明書を必ず読み、記載の手順に従ってご使用ください。
お読みになった後は、お使いになる方がいつでも見ることができる場所に、大切に保管してください。
●第三者に譲渡・貸与される場合も、この説明書を必ず添付してください。
●当商品に関するお問い合わせは、お買い求めの販売店もしくは当社にご連絡ください。

取扱説明書をよく読み、指示に従う。

この説明書には拡大損害が予想され、禁止事項には必ずお守り頂きたい強制事項に はマークを付けて表示しています。






:',63

⑨ 外枠


⑩ 標準裏蓋





② 長針





PP-(:(/('
6+2&.3522)





③ 長針用
目盛板
⑦ スピンドル
⑧ 測定子

④ 短針



⑤ 短針用目盛板⑥ ステム①リミット針 …… 合否判定を素早く行うための、
合格範囲の目印です。
②長針 …………… 目量を表す針です。
③長針用目盛板 … 目量を表す目盛です。⑨外枠を回すと回転します。
④短針 …………… 目量×１００を表す針です。
⑤短針用目盛板 … 目量×１００を表す目盛です。
⑥ステム ………… 保持具取付部です。

⑬ 耳金付裏蓋

保持具









:',63










PP⑥ ステム

保持具への取り付けは、ステムまたは耳金のみで行う。



PP-(:(/('
6+2&.3522)







支柱

なるべく短く・ステム、耳金以外での取り付けは、精度不具合や製品の破損の原因となります。

取り付けの際、ステムを締め付けすぎない。

・ステムを締め付ける力が強すぎると、スピンドル動作不良の原因となります。

※保持例

・測定室の温度変化や、ダイヤルゲージの繰返し誤差が原因として考えられます。
下記の解決策をお試しください。
●測定室の温度が一定となるよう、環境を整える●測定時に任意の一定周期毎にマスタゲージを使用して、温度変化や繰返し誤差から来る原点ズレを調整する

ステムまたは耳金部で保持具に取り付けてください。
他の箇所での取り付けは、精度不具合や製品の破損の原因となります。

３．長針・短針・スピンドルの動きが滑らかであることを確認する

指の腹で測定子をゆっくり押して上下させ、動きを確認してください。
滑らかでない場合は、ステムを締め付けすぎている恐れがありますので、調整してください。
また、針が設定位置で安定して静止することを確認してください。

４．測定対象に対して、スピンドル（測定子）が垂直に接するよう設置する■測定値が安定しない、測定精度外の数値を示す

・測定子の磨耗が考えられます。
測定子が磨耗すると測定値に影響が出ます。定期的に磨耗確認を行い、測定値に影響が出る場合は交換を行ってください。

本体仕様
直角

長 針 1/10回転 全測定範囲 繰返し
目量 測定範囲
質量目盛仕様 戻り誤差 測定力 １回転 指示誤差 指示誤差 精密度（mm） （mm）（μm） （N） （mm） （μm） （μm） （μm）（平蓋装着時）
210 J
WDI-3058SP
30
7
3.0 以下
14
35
5
270 J
WDI-5058SP
50
93.5 以下15
40
0.01
0 -1001
WDI-8058SP
290 J
80
11
4.0 以下―50
6
WDI-10078SP
530 J
100
15
4.5 以下
―
品

測定対象

測定対象

使用方法（比較測定）例）合格範囲：−０.０５〜 ＋０.１５mm① リミット針番●交換用測定子品番：DI-CP

使用後のお手入れ・保管方法

２．原点を設定する

使用中に付着した埃、汚れを清掃する。※注油は行わないこと。

保持位置を上下させるか、外枠を回して長針用目
盛板を回転させ、長針を「０」に合わせます。



② 長針

※リミット針について










:',63

PP・スピンドル摺動面の汚れは、乾いた布かアルコールを少量含ませた布で拭き取ってください。
・その他外観の汚れは、柔らかく乾いた布か、中性洗剤を少量含ませた布で拭き取ってください。



３．マスターを外して、測定する

PP-(:(/('
6+2&.3522)











リミット針は自由に動かして、合格範囲の目安として使うことができます。

使用方法（平行度・平面度・偏心測定等）
１．測定子を測定箇所に当てる

測定子を指の腹で軽くつまんで持ち上げて、上から降ろすように測定箇所に当てます。測定子の側面から衝突させないように注意してください。

２．原点を設定する

保持位置を上下させるか、外枠を回して長針用目盛板を回転させ、長針を「０」に合わせます。

３．測定対象を動かし、目盛を読み取る

保持具のたわみ・浮きに注意する。

・保持具は十分に剛性のあるものを使用し、ダイヤルゲージの重みによる保持具のたわみを防いでください。
・保持具はダイヤルゲージの測定力に対して、十分に固定可能なものを使用し、測定時の保持具の浮きを防いでください。
・保持具の支柱はなるべく短く設定し、支柱のたわみを防いでください。
※たわみ、浮きは原点位置ズレや指示値のバラつきなど測定
誤差の原因となります■比較測定時、使用前と使用後で原点の位置がズレる

２．保持具にダイヤルゲージを取り付ける

スピンドルに衝撃を与えないようマスターを静かに外し、測定物を当てて目盛を読み取ってください。

横方向へ力を加えないトラブルシューティング

ゆるみがある場合は、裏蓋の全てのねじを締め直してください。

測定子を指の腹で軽くつまんで持ち上げて、上から降ろすようにマスターに当てます。測定子の側面から衝突させないように注意してください。

急激に
動かさない

ダイヤルゲージはマグネットベースやコンパレータスタンドなどの保持具で保持して使用します。
取り付けの際は下記の内容をお守りください。

⑦スピンドル …… 測定時に上下する軸です。
⑧測定子 ………… マスターやワークに当てる測定部です。
⑨外枠 …………… ③長針用目盛板を回転させます。
⑩標準裏蓋 ……… フラットなタイプの裏蓋です。
⑪耳金 …………… 保持具取付用パーツです。
⑫取付穴 ………… 保持具取付部です。
⑬耳金付裏蓋 …… 耳金が付いた裏蓋です。

１．測定子と裏蓋にゆるみが無いことを確認する

１．マスターを設置する

ご使用の前に※耳金付裏蓋を使用する場合は、標準裏蓋のねじを全て外し、
耳金付裏蓋と交換してねじで取り付けてください。

使用方法（準備）

スピンドル（測定子）が垂直に接しないと、測定
値や動作に不具合が出ます。
常に垂直を保つように設置してください。
※フライスのバイス平行確認に使用する際は、バイス測定面とスピンドル（測定子）の垂直を保ちつつ、フライス盤のテーブル送り時に支障とならないように、マグネットベースの位置を調整してください。

スピンドルに衝撃を与えない。

・落下などの衝撃を与えたり、重量物を乗せたりしないでください。精度不具合や故障、破損の原因となります。

⑪ 耳金

⑫ 取付穴
（ȭ６．
５ｍｍ）

※WDI-3058SP・WDI-5058SP・
WDI-8058SP………ȭ 8ｍｍ※WDI-10078SP ……ȭ10mm

・スピンドルを急激に動かしたり、横方向への力を加えないでください。精度
不具合や故障、破損の原因となります。

本器は大切に取り扱う。

[付属品]

PP

・指定用途以外へのご使用は、製品の破損や磨耗、予測できない事故の原因となります。
●温度０〜４０℃、湿度３０〜７０％の結露しない場所●埃、油、オイルミストが少なく、直射日光が当たらない場所●子どもや、使用者以外が近付かない場所
・上記に反する場所での使用は、精度不良や製品の破損、事故やけがの原因となります。



・精度不具合や製品の破損の原因となります。

下記の条件を満たす環境で使用する。

① リミット針

分解、改造を行わない。

・取扱説明書に記載された内容以外での使用は、事故の原因となります。

精密比較測定作業にのみ使用する。

、■各部の名称と機能 ※イラストはWDI-3058SPです。



してはいけない内容『禁止事項』です。

注意

安全上のお知らせる事項には必ずお守りください。

■表示内容を無視して、誤った使い方をした時に生じる危害や損害の程度を次の表示で区分しています。

この度は「ロングストロークダイヤルゲージ」をお買上げ頂きありがとうございます。この商品はマグネットベースやコンパレータスタンドなどの保持具に固定して、任意の基準地点を０とした時の測定値の増減を読み取る比較測定器です。

■用途の一例 ・マスターを基準とした加工物の検査

安全上のご注意

定対象をゆっくりと動かし、針の動きを読み取ってください。

測定子の磨耗を確認する。

・測定子が磨耗すると測定値に影響が出ます。定期的に磨耗確認を行い、測定値に影響が出る場合
は交換を行ってください。

付属のケースに収納し、乾燥した冷暗所に保管する。

・保管時、スピンドルに負荷をかけないよう、力（押し込み力・曲げ荷重など）をかけないでくだ
さい。
・直射日光や湿気は避け、管理者以外が触れない状態で保管してください。

校正について●測定精度を保つため、一定周期で校正による精度確認をおすすめします。
使用を繰り返す間に測定面の磨耗などにより、精度を保てなくなる可能性があります。定期的な精度確認の実施をおすすめします。

弊社にて校正を受け付けております。
全国にある当社代理店、取扱店を経由してお申し込みください。

お問い合わせ：（０２５６）３３-５５０２

発売元〒955-0055 新潟県三条市塚野目5丁目3番14号(0256)33-5502(代) FAX(0256)33-5528

URL http://www.niigataseiki.co.jp

I 370-K1 2205