MULTI FUNCTION INSTALLATION TESTERKEW 6514BT
Versatile tester supporting various electrical testing for check,
maintenance and troubleshooting of electrical installations.
The EVSE dedicated test function offers
advanced diagnostic analysis when used with EVSE adapter.
(EVSE: Electric Vehicle Supply Equipment)

Electrical tests
Insulation

Earth
RCD

2-Wire LoopFrequency

AC VContinuity

Motor rotation

Phase rotation
SPD(Varistor)

Tests dedicated for EVSE
Waveform checkof CP signal

Latch Locking

Programmable
Auto testOffers various tests for electrical installation
Loop 2-Wire
KEW 6514BT has an advanced measurement method called “Loop 2-Wire”.
It can carry out Loop impedance test by using 2 wires only,
while using 3 wires is usually required.
Its small test current of 7mA does not trip even the 15mA RCDs!

Motor rotation check
Without power supply, the motor connection can be checked by electromotive force
generated by spinning the three-phase motor axis (shaft) by hand or other means.

Safety lock function for insulation resistance range
This safety function prevents the unintentional voltage presence for insulation resistance range.
This safety function can be set on each range by our special application.

Auto memory
Activating this function, KEW 6514BT automatic saves measured data at
every test. Up to 1000 measured data can be saved in the internal memory.
Test can be carried out by pressing the test switch on KEW 6514BT or
the switch on the remote test lead. Pressing and turning the test switch on
the instrument, you can have continuous test operating.

USB & Bluetooth®

Measured data saved in the internal memory canbe transferred to PC via USB adapter (8212-USB)
or Bluetooth® and it is possible to manage themeasured data and print complete test report.
* 8212-USB is an optional accessory.
* Please download the software from our website.

Color LCD Display
All the test data is shown in one large
colored screen.
HELP function will show how toconnect the instrument according to
the function selected.

Measurement DataManagement app"KEW Smart Advanced"
application facilitates the check ofany measured result on tablet or
smartphone via Bluetooth®.
User template (xlsx format) can be
uploaded to the device andthe measured values areautomatically entered. Using thisapplication, KEW 6514BT allows toactivate or not each function andeasily change the settings such ascomparator value for each function.
Bluetooth®

Dedicated function for EVSE

Offers advanced diagnostic analysis
when used with optional EVSE ADAPTER

CP signal analysis function
Connecting KEW 6514BT tothe CP terminal of theEVSE ADAPTER, CP signal
can be deeply analysed asusing an oscilloscope.
Also CP STATE and charge
current are automaticallycalculated and shown on theinstrument display.

KEW8601/8602
EVSE
KEW 6514BTProgrammable Auto test

Resistance measurementof latch switch circuit

There is an EVSE dedicated function where you cancarry out various tests. Combination and sequence oftests can be customized by using our special app.
The instrument supports you showing how to makethe connections on its screen prior to test and also
gives a step-by-step guide to carry out all thenecessary tests.

Using KEW 6514BT in combination with EVSE ADAPTER
KEW 8601, you can measure the circuit resistance andcheck if the latch switch of EVSE with type1 connector
operates properly, with PASS or FAIL results.

KEW 6514BT comes complete with everything you needfor testing an electrical installation● MULTI FUNCTION TESTERAccessories

Selection Guide

KEW 6514BT

VOLT✓

EarthOptional accessories
MODEL 7272

KEW 6516BT

600V

Continuity

RCD

KEW 6516

2-wire / 3-wire
Function

x1/2, x1, Ramp

RCD typeAC(G)x1/2, x1, x5, Ramp, Auto, Uc
AC/A/F/B(G/S)

Test current

15/30/50/100/
200/500mA

10/30/100/300/500/1000mA/variable

PSC/PFC-✓EVSE RCD (Compatible with RCD for EVSE)
Test
LOOP (Compatible with RCD for EVSE)-✓(DC 6mA)-✓(N-PE:4mA)

Checking CP signal✓-Latch switch circuit resistance✓-

Programmable Autotest✓-

25/50/100/125/250/500/1000V

100/250/500/1000V

Precision measurement cord set
2 cord reels with test leads,
2 spikes, an earth test lead,a carrying case.

Insulation
resistance

Test Voltage

MODEL 8259

MODEL 8212-USB

Loop Impedance

Adapter formeasurement terminal[red, yellow, green/1set]

USB adapter

Loop ATT 2-wire✓(L-PE:7mA)✓(L-PE:15mA)

Loop ATT 3-wire-✓

Loop HIGH-

KEW 8601

KEW 8602

EVSE ADAPTER
TYPE1 plug

EVSE ADAPTER
TYPE2 plug

Motor rotation check✓-✓

Auto memory
Communicationinterface✓✓

Phase rotation

Bluetooth®

✓USB✓-✓

● MULTI FUNCTION TESTER KEW 6514BT Specification
Insulation resistance
25V
50V
100V 125V
250V
Test voltage
2.000/20.00MΩ
2.000/20.00/200.0MΩ
Range(Auto-ranging)(Auto-ranging)
AccuracyFirst effective
measurement range
Second effective
measurement range

Rated current
Short circuit current
RCD
Rated voltage

0.100 to 10.00MΩ 0.100 to 25.0MΩ
±2%rdg±2dgt
0.050 to 0.099MΩ
±2%rdg±4dgt
10.01 to 18.00MΩ 25.1 to 180.0MΩ
±5%rdg
1.0 to 1.2mA
1.0 to 1.2mA@0.025MΩ(25V)@0.1MΩ(100V)@0.05MΩ(50V)@0.125MΩ(125V)
1.5mA max
85 to 440V(50/60Hz)
×1/2, ×1, Ramp

Function

15/30/50/100/200/500mAAC(G)RCD type
Accuracy Trip
currentx1/2x1
Ramp
Trip time x1/2x1

Continuity
Range
Open circuit voltage (DC)
Measurement current
Accuracy
Earth
Range
Accuracy
Volts
Range
Measuring Volts
range
Frequency
Accuracy Volts
Frequency
GeneralApplicable standardsCommunication interface
Power source
Dimension
WeightAccessories

Optional accessories-8 to -2%
+2 to +8%
-4 to +4%
±1%rdg±2ms
20.00/200.0/2000Ω (Auto-ranging)
7 to 14V
200mA or more(2Ω or less)±2.0%rdg±8dgt
20.00/200.0/2000Ω(Auto-ranging)±2%rdg±0.08Ω(20.00Ω)±2%rdg±3dgt(200.0/2000Ω)500V
2.000/20.00/200.0/1000MΩ
(Auto-ranging)–
0.100 to 50.0MΩ 0.100 to 100.0MΩ

1000V
2.000/20.00/
200.0/2000MΩ
(Auto-ranging)
0.100 to 1000MΩ

50.1 to 180.0MΩ 100.1 to 900MΩ

1001 to 1800MΩ

1.0 to 1.2mA@0.25MΩ

1.0 to 1.2mA@1MΩ

1.0 to 1.2mA@0.5MΩ

EVSE Function
Measuring Vtop
range
Vbase
Frequency
Duty
Charging current
Accuracy Vtop
Vbase
Frequency
Duty
Charging current
Loop impedance(L-PE(2-wire))
Rated voltage
Impedance range
Accuracy
Measurement current
Phase rotation
Measuring Phase rotation
range
Motor rotation
Indication

300.0/600V(Auto-ranging)
2 to 600V
45 to 65Hz
±2%rdg±4dgt±0.5%rdg±2dgt

SPD(Varistor)
1000V max.
0 to 1000V(goes up by 1V)±5%rdg±5dgt––––

2.0 to 15.0V-15.0 to -2.0V
980 to 1020Hz
10.0 to 96.0%
6.0 to 80.0A±4dgt±0.5%rdg±4dgt±10dgtDepends on the accuracyof the Duty cycle
85 to 260V(50/60Hz)
200.0/2000Ω
±3%rdg±10dgt
L-PE:7mA
3 to 600V(45 to 65 Hz)
0.1 to 2V(1 to 10 Hz)
Clockwise direction:
1.2.3 and clockwise phasesequence icon
Counterclockwise direction:
3.2.1 and counterclockwise
phase sequence icon

IEC 61010-1, IEC 61010-2-030 CAT Ⅳ 300V / CAT Ⅲ 600V Pollution degree 2, IEC 61010-2-034
IEC 61557-1,2,3,4,5,6,7,10, IEC 60529(IP40)
USB, Bluetooth® 5.0*1
LR6(AA)(1.5V) × 8136(L) × 235(W) × 114(D)mm
Approx.1300g (including batteries)
7281(Test leads with remote control switch), 7247(Distribution board test lead)
7228A(Earth resistance test leads), 8041(Auxiliary earth spikes[2spikes/1set])
8017B(Extension prod long), 8923(Fuse[0.5A/600V]) × 1 (included), 1 (spare), 9084(Soft case)9142(Carrying case),9151(Shoulder strap), 9199(Shoulder pad), Batteries, Instruction manual8259(Adapter for measurement terminal), 7272(Precision measurement cord set)
8212-USB (USB adapter), 8601(EVSE ADAPTER), 8602(EVSE ADAPTER)*1 Some countries regulate the compliance with their Radio Law of the products equipped with Bluetooth®.
Please confirm it with your distributor before purchasing our products equipped with Bluetooth®.
*Bluetooth® is a trademark or registered trademark of Bluetooth SIG. Inc.
*Android is a trademark or registered trademark of Google LLC.
*iOS is a trademark or registered trademark of Cisco Technology, Inc. in the United States and other countries.

For inquiries or orders :●This leaflet was produced in October 2024. The contents of this leaflet are subject to change without notice.

LC_6514BT_3_0_Oct24_SS_en