INSTRUCTION MANUALDIGITAL S-LINE INDICATOR

Model No. DEI-WKS-N

Thank you for purchasing the Digital S-Line Indicator.
For proper operation, please read this manual thoroughly before use.

PARTS IDENTIFICATIONFEATURES● Tolerance Setting capability

LCD Display

Display Rotates 270°
Easy to read from any position.

● Calculation capability● Display rotates 270°

● Programmable “Favorite” Button

DEI-WKS-NSWISS MADE

(Can be set for one touch access to frequently used functions.)● MIN、MAX、MINMAX (TIR) display functions● Direction of measurement (＋ / −) switchable● Preset function● Sleep mode (after approx. 20 min.)[MODE] Button

MODE[SET] Button

SETBattery CoverCenter (“Favorite”)
ButtonAccessories

Stem (Mounting Shaft)

ＣＲ２０３２

Contact Point

Measurement Range

0.001mm/0.0001mm(switchable)

2.3μm
Less than 1.5N
0.5μm
φ8mm
5〜40℃
CR2032 (Lithium cell) ×1
140g

Accuracy
Measurement ForceBATTERY REPLACEMENT①Remove the (2x) battery screws using (+) PH#0 screwdriver.
Gently pry open the cover using a small standard screwdriverin the notch and slide out the battery.
②Insert the new battery following the polarity markings on thecover.
③Replace Battery Cover and tighten screws to secure.

Repeatability
Mounting Shaft Diameter
Operating Temp. Range
Power SupplyWeight

TROUBLESHOOTING
Error Condition

Battery

Battery Cover＋

Measured value is wrong

Clean measuring surfaces withlens tissue or similar.
● Confirm and re-set 0-Point setting.

●●LCD does not display
Displayed value isunstable● Display is hard to readNotch●

NOTICE Use only CR2032 type battery.

POWER【Power ON】
Press [MODE], [SET], or Center Button to turn on power.
【Power OFF, Sleep Mode】
Press and Hold the [SET] button to enter Sleep Mode; continuepressing to turn power OFF.
1mm

Corrective Action●ＣＲ２０３２[SET] Button (Press and Hold)

DEI-WKS-N
0-12.5mm
0.0001mm

Model No.

Displayed Resolution※Included for test

[SET] Button (Press and Hold)

SPECIFICATIONSResolution

SpindleCR2032 (Lithium Cell)

Battery Cover● Made in Switzerlandmm

Sleep Mode
Power OFF※Sleep Mode is entered automatically if gauge is idles for about
20 min.
※To exit Sleep Mode, press any button or move the Spindle.
※When remaining battery level is low, the『 B 』icon is displayed;
please replace battery.
※To reset all settings, press and hold [SET] and [MODE] atsame time for about 4 sec., “” will display.

Cycle power by removing andreplacing battery.
● Replace battery with new one.

※If problem persists, or if you have any questions, please contact distributor
or place of purchase.
※Please note, manufacturer is unable to respond to inquires or provide
service directly. Please contact distributor or place of purchase.

HOW TO USE①Mount indicator securely, such as on optional Magnetic Base,and make sure that the Spindle is perpendicular to themeasurement surface.
②Press [MODE], [SET], or Center Button to turn on power.

③Check that the probe is in contact with surface, and refer to thefollowing to set the 0-Point.
Refer toRefer toFUNCTION SETUP: “Preset Value”

Programmable “Favorite” Button: “Zero setting”

LCD will display “0.000mm” or “0.0000mm”.

④With Gauge body held fixed, and the probe in contact with theworkpiece, the LCD will display the difference between
workpiece and reference plane.
See Reverse for Feature Settings →FUNCTION SETUP【Measurement Display】
Press the [MODE] Button, or Press-and-hold, to access and cycle through the functions
as show in the chart below. When the function is displayed, press Center Button to access.
※If no operation is performed after 5 seconds, display will return to measurement screen.

1mm※ This option
not used[MODE] Buttonmm[MODE] Button[MODE] Button

Long Press【REF1 / REF2 Mode】
Two different Zero points, REF1 and REF2 canbe used (also Tolerance settings.) This feature ishelpful when there are multiple measurement
criteria in use at the same time.

1mm[MODE] Buttonmm

Press Center Button to display REF1 / REF2, thenpress [SET] to select.

→Display maximum displacement without regard to direction (TIR).

[MODE] Button

To SetPress Center Button to display options, then press[SET] to select.

Return toMeasurement Display

Center Button

1mm[MODE] Button
1mm[MODE] Button●Center Button can be programmed to call up a frequently usedfunction with a single press while taking measurements.
※Programmed setting is reset when power is turned off.
To SetmmTolerance
Setting function
ON / OFF[MODE] Button[MODE] Button(−), thenPress Center Button to move the cursor, and [SET]
to change the value of that character. Press [Mode]to enter.
Ex: If computation factor is set to “2.00", and measured value is “4.25”(actual amount of Spindle motion,) the displayed reading will be “8.50”※Set to 1.000
to disable【Auto Off】

mmTo Set

Press Center Button to display “and press [SET] to select.

CAUTION● This is a precision instrument, handle with care. Do not drop or subject

to excessive forces.
● Keep away form high voltage and strong magnetic fields; instrument
damage may result.
● Do not use or store in direct sunlight or locations with hightemperatures such as in a car, or near a stove or heat source.
● When taking measurements, hold gauge securely, such as inmagnetic base. (sold separately)● Please do not disassemble or modify.
● Remove dust and cutting chips after use and apply rust preventative
oil to metallic surfaces. Please keep out of reach of children.
● Do not use organic solvents to clean the body of the instrument.
● When not being used for a long period of time, please remove battery.
● Only use as directed. Improper use may cause accident or injury.

(＋) /To Set

Calculation
function
Center Button

Press Center Button to display
press [SET] to select.

【Calculation Function】
You can enter a number to multiply by theamount of Spindle movement.

mmCenter Buttonmm

Red Light

Press Center Button to display 0.001 / 0.0001,
then press [SET] to select.

To SetMeasurement
Direction ＋ / −mm

OK Green Light【Measurement Direction】

mmCenter Button

SwitchREF1 / REF2

Orange Light

To Set[MODE] ButtonResolution setting
0.001 / 0.0001

mm

Center Button

Center Button【Resolution】

mm

NOR, MIN,MAX, DELTAmodemm

Center Buttonmm

Orange Light

Center Button※This option
not used

OK Green LightTo reverse the display indication, reverse the entry of the Lower andUpper limits.

Center Button

Zero settingmm

Red Light

Presetmm

Center Buttonmm※Tolerance values can be set for MIN,
【Tolerance Function】
MAX, and DELTA modes.
With tolerance values set, gauge will show if
measurement is within range.
Colored lights will show tolerance determinationfor easy judgment.

Measurement: 2.25mm Measurement: 4.00mm Measurement: 7.25mm
mm

Center Button

Reset MIN,MAX, DELTA
displayTOL INPUT

Ex: For measurement Lower limit set to “+6”, Upper limit set to “+3”Center Button

Hold function

Set UpperLimit

Press Center Button
to move the cursor,and [SET] to change
the value of thatcharacter. Press[Mode] to enter.

Measurement: 2.25mm Measurement: 4.00mm Measurement: 7.25mm

Long Press

Center Buttonmm

TOL

Ex: For measurement Lower limit set to “+3”, Upper limit set to “+6”Press and hold the Center “Favorite” Button to enter the button
programming selection mode.Press Center Button to cycle through theoptions and then press [MODE] or [SET] to select.

※This option
not used

Set Lower
Limit
TOL INPUT

Programmable “Favorite” ButtonmmTOL

Center Button[MODE] Button

Press Center Button for numeric display. Press
Center Button to move the cursor, and [SET] tochange the value of that character. Press [Mode] toenter.
●Press [SET] in measurement screen to set display
the preset value.

Center Button

1mm

1mm

1mm

To Set[MODE] Button(MAX),※“deltA” (MINMAX)To Set【Preset Value】
Any number can be preset for use in setting display
PRESET INPUT during measurement by using this function.

1mm【Display Mode】(MIN),
Set measurement mode:(MINMAX),or(NORMAL)[MODE] Button『』…disabled 『”/“”,』…Auto off if idle for 20 min.

【Button Lock】
For locking buttons while taking measurements.

mm

To Set[MODE] Button

Press Center Button to display ON / OFF, thenpress [SET] to select.
●When lock is ON,『』icon will display on LCD.
●To release the lock mode, press and hold [SET]
until “” is displayed.

Niigata seiki Co., Ltd.
6-15-22, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-31-5670 Fax. : +81-256-39-7730

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

I303-K1606