User Manual

DIGITAL HOLE PITCH CALIPERS●For safe and proper use of this product, please read this instruction manual before use and follow the procedures
described. Keep this manual with the calipers for future reference.
●Keep this manual with the instrument if transferred or leased to a third party.
●For inquiries about this product, please contact dealer or Niigata Seiki.

■PART IDENTIFICATION AND FUNCTION④Lock Screw●GDCP-200TA⑤ [HOLD] Button⑥ Output Connector ⑩ Scale② Jaw⑧ [ZERO/INC] Button⑨ [ON/OFF/P] Button

SAFETY PRECAUTIONS Please ObserveAlways follow these marked operating procedures in order to prevent harm
to yourself or others, and to prevent damage to property.

WARNING

Denotes a requirement You MUST do

Indicates risk of personal injury or property
damage if not followed.

Read this manual and follow the instructions.

・Use of product other than as described in the manual may cause accident.
Use only for measuring.
・Use for any purpose other than measuring may cause damage orwear to the instrument. Improper use may also cause accident.

Handle with Care.

・Do not drop or subject instrument to excessive shock. Do not place underheavy objects. Improper handling may cause damage or poor accuracy.
・Do not scratch instrument.

Avoid using in locations with high electric fields, such as nearfluorescent lights or switching power supplies.
・May cause damage to Display Unit or erroneous reading.

Use and store only in environment meeting the following conditions:●Dry location protected from rain, water, and oils.
●Location that does not become hot.
●Location protected from use by children and unauthorized people.
・Use in places contrary to the above may cause damage to the productresulting in accident or injury.

Jaw Tips are Sharp̶Handle With Care.
・Careless handling may cause injury.

Do not disassemble or modify

・Doing so may cause accident or injury. Please contact distributor orplace of purchase for service.

Do not write on instrument with electric engraver.

・Do not engrave instrument body or jaw, such as an ID number withengraver or electric pen.

Battery
Cover■Setting Preset for Reference Plate Dimensions●Terminals may be broken if battery ispressed straight down.
●Insert battery at angle, underneath the small hook, as shown.
Battery●GDCP-200AA⑪ Reference PlateAB① Measurement Surfaces① Measurement Surfaces … Surfaces which contact the workpiece.
②Jaw ……………………… For positioning on object to be measured.
③LCD Display ……………… Displays digital measurement.
④Lock Screw ……………… Tighten to prevent the slide and display from moving.
⑤ [HOLD] Button …………… When pressed, "HOLD" Icon appears on Display
and measurement will be fixed.
⑥ Output Connector ……… Connect to optional BluetoothBox to sendmeasurements to PC.

①To turn on, press the [ON/OFF/P] Button.
②To turn off, press and hold (>2 sec.) the [ON/OFF/P] Button.
③With power on, press the [HOLD] Button to fix the value onthe display.

HookSlide InTerminals⑦Battery Cover

Denotes a prohibition You MUST NOT do●Use care to avoid damaging
terminals when inserting battery.

（For Hole Pitch Measurement）②Jaw① Measurement
Surfaces

Use only CR2032 typebattery.

Pressed Down

（For Hole Center toEnd Face Measurements）③ LCD Display

NoticeTerminals

Battery

TROUBLESHOOTINGTROUBLESHOOTING
Q. Nothing is displayed on LCD.

⑦Battery Cover ………… Open to access battery.
⑧ [ZERO/INC] Button … When pressed, Display will read "0.00".
⑨ [ON/OFF/P] Button … For turning power ON/OFF, and for
Preset Function.
⑩Scale ………………… Supports LCD Display Unit.
⑪Reference Plate ……… For setting origin using the Preset
Function. Includes Inspection Report.

A. The battery may be running low, please replace battery
with a new one.

Q. With power on, the number on the LCD does notchange as Display Unit is moved.
A. Please reset by removing battery for about 30sec., andthen replacing.

Q. Nothing happens when buttons are pressed.
A. Make sure battery is properly inserted.

Q. The characters “MIN”, “MAX”, or “△” are displayed onLCD.

PREPARATION FOR USE
B efore using, please follow the steps below:●Wipe off oil, rust proofing, and dirt from measuring
surfaces and slide.
Contamination of surfaces may cause measurement
error.
●Allow the caliper and object to be measured time toreach the same temperature.
A temperature difference between the object to bemeasured and the caliper may cause measurement error.
Allow enough time for the temperatures to equilibrate.

PRODUCT SPECIFICATIONS
Model No.

※Not including quantization error (±1 count)
Maximum
Measured
Permissible Weight
Hole Diam.
Error※

Measurement
Range

GDCP-200AA 20-200mm
300 g
䃥1.5〜19.5mm ±0.05mm
GDCP-200TA 10-200mm
260g●Resolution . . . . . . . . . . . . . . 0.01mm●Display . . . . . . . . . . . . . . . . . 6 Digit LCD●Auto Off . . . . . . . . . . . . . . . . approx. 20 min●Max. response speed . . . . Slide to Right.....1.5m/s
Slide to Left......0.75m/s※Left/Right as shown in diagram at topof manual.

C〜40°
C●Operating Temp. Range . . 0°
C〜 60°C●Storage Temp. Range . . . −0°●Power . . . . . . . . . . . . . . . . . . (1x) CR2032 Lithium Battery●Battery Life . . . . . . . . . . . . . . approx. 2.5 years in normal use.
●Accessories . . . . . . . . . . . . . ・CR2032 Lithium Battery(1x, for test)
・Reference Plate

HOW TO USE■Power ON/OFF①Slide cover in direction of arrow.
②Insert battery with (+) side outward.
③Replace battery cover.

vemoThank you for purchasing Niigata Seiki Digital Hole Pitch Calipers.
This precision measuring tool measures distance between holes, or distance from hole center point to end face.

INSERTING・REPLACING BATTERYRe

HOLE CENTER POSITION
DIGITAL CALIPERS

Model No. GDCP-200AA /GDCP-200TA

A. Cycle power, or apply several long presses to [HOLD]
Button. Use as normal once character is off.

Q. The correct value is not displayed.
A. High humidity may cause dampness. Please dry off theinstrument and use only in dry location.

SERVICE●If Instrument is not working properly, or if you have anyquestions, please contact distributor.
●Please note, manufacturer is unable to respond to inquires
or provide service directly. Please contact distributor.

AFTER USE CARE・STORAGE●After use, wipe off any dirt or oil from measuring
surfaces, sliding surfaces, and exterior using a dry
cloth and perform rust prevention treatment.
For rust prevention, lightly apply anti-rust oil, or place in rustpreventive bag.

●Store with Lock Screw loosened.

If stored with Lock Screw tight, expansion due to temperaturechanges may cause excessive force which will cause
inaccuracy.

●Store in a cool, dry, and dark location in the provided
case.
Keep out of direct sunlight and moisture, and please keep
secure from unauthorized personnel.

●Remove battery if Caliper will not be used for a long
period of time.
Batteries may leak if left in unused instrument for an extended
period.

※It is necessary to set the Preset Value to the dimensions
specified on the Reference Plate Inspection Report beforefirst use of product, and after replacing the Battery.
※Preset Value is stored in memory even when power is
off so there is no need to set with each use.

The Reference Plate Inspection Report, will be required for thisprocedure, please prepare.
①Press and release the [ON/OFF/P] Button, “P” icon willappear and flash on the Display.
②With the “P” icon displayed, press and hold the [ON/OFF/P]
Button to change the position of the blinking character.
③When the desired character is blinking, a short press of the[ON/OFF/P] Button changes the number at that position.
④ Repeat steps ②,③ to set the value to the dimension printed
on the Inspection Report that came with the Reference Plate.
※The dimensions listed on the Inspection Report are accurate
to three decimal places. Please round off the number whenentering into the display.
⑤Move the blinking position back to the “P” icon and press the[ON/OFF/P] Button again briefly. The entered value is now preset.

■Setting Reference Point※Please set the Origin before each use.

Using the provided Reference Plate, set the Origin before
taking measurements.
①Prepare the Gauge and Reference Plate by cleaning themeasurement surfaces with a cleaning solution such as aparts cleaner.
②Turn on power, and loosen the Lock Screw.
③Position the Caliper to measure the Reference Plate,（Fig.1,Fig.2） and apply two short presses to the [ON/OFF/P]
Button.（Fig.3）④Confirm that the reading on the LCD matches the Inspection
Report for the Reference Plate. The Origin is now set and thegauge is ready to start measurements.

（Fig.2）●GDCP-200AA
１
Measure（Fig.1） Reference PlateAB

GDCP-200TA GDCP-200AA
MeasureMeasureReference Plate（Sectional view）●GDCP-200TA（Fig. 3）●GDCP-200AA,& 200TA

１ Measure
２ Press Twice

Reference Plate（Sectional view）

Niigata seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

取扱説明書品番

GDCP-200AA / GDCP-200TA

穴ピッチ用デジタルノギス

穴の芯間測定用
デジタルノギス● GDCP-200TA（端面から穴芯までの測定用）⑥出力コネクタ②ジョウ⑩ 本尺● GDCP-200AA（穴の芯間測定用）

② ジョウ

①測定面A

B

端子③LCD表示部 ……… 測定数値をデジタル表示します。

⑨ON/OFF/Pスイッチ …電源のON/OFF、プリセット数値の設定と表示を行います。

②ジョウ …………… 被測定物を挟んで測ります。

④止めねじ ………… デジタル液晶部のスライドを固定します。

⑤HOLDスイッチ … 押すと画面上に"HOLD"と表示され、表示している数値を固定できます。
⑥出力コネクタ …… 別売の BluetoothBOX を接続することで、測定値をパソコンへ出力することができます。

⑧ZERO/INCスイッチ … 押すと画面に"0.00"が表示されます。

⑩本尺 …………………… デジタル液晶部のスライドを保持します。

⑪マスタプレート ……… プリセット機能を使い、原点を設定するためのマスタです。検査成績表付。

必ずお守りください

お使いになる人や他の人への危害、財産への損害を未然に防止するため、必ずお守りいただくことを下記のように説明しています。
必ず実行していただく内容『強制事項』です。

注意 するおそれがある内容』です。

お守りいただかないと『傷害を負う、または物的損害が発生取扱説明書をよく読み、指示に従う。

・取扱説明書に記載された内容以外での使用は、事故の原因となります。

測定作業にのみ使用する。

・指定用途以外へのご使用は、製品の破損や摩耗、予測できない事故の原因となります。

大切に取り扱う。

・ぶつける、落とすなどの衝撃を与えたり、重量物を上に載せたりしないでください。精度不良や破損の原因となります。
・商品に傷をつけないでください。

インバータ式蛍光灯などの電界が発生する場所での使用は避ける。

・ＬＣＤ表示部の数値が誤作動を起こす原因となります。

下記の条件を満たす環境で使用・保管する。

●雨や水、油などがかからない、乾燥した場所●高温にならない場所●子どもや、使用者以外が近付かない場所
・上記に反する場所での使用は製品の破損、事故やけがの原因となります。

ジョウ先端は尖っているので、扱いに注意する。

・けがの原因となります。

分解・改造を行わない。

・製品の破損や事故の原因となります。修理はお買い上げの販売店、または当社までご連絡ください。

電気ペンなどで書き込まない。

電池

・電気ペンなどで本尺やジョウなどに、番号などの書き込みをしないでください。

ご使用前に、下記の準備を行ってください●本体と被測定物の温度慣らしを十分に行う測定器本体と被測定物の間に温度差があると、誤差の原因となります。
同じ温度になるよう、温度慣らしは十分に行ってください。

品番

測定範囲（mm）※量子化誤差（±1カウント）含まず

適用穴径（mm）

GDCP-200AA 20-200
䃥1.5 〜19.5
GDCP-200TA 10-200

Ｑ．電源ONの状態でスライドさせてもLCD表示の数字が変わらない。
Ａ．一度電池を取り外し、30秒後に電池をセットし直してください。

Ｑ．LCD表示部に『MAX』『MIN』『△』などの文字が表示されている。
Ａ．一度電源を切るか、”HOLD”スイッチを何回か長押しして文字を消してからご使用ください。

●測定面、摺動面、外観などの汚れや防錆油などを拭き取るそのまま測定を行うと誤差の原因となります。

製品仕様

Ｑ．LCD表示部に何も表示されない。
Ａ．電池の残量が少なくなっている可能性があります。新しい電池と交換してください。

Ｑ．スイッチを押しても何も反応しない。
Ａ．電池が正しくセットされているか確認してください。

使用前の準備

最大許容誤差 質量（mm） ※ （g）±0.05300260●最 小 読 取 値：0.01mm
示：LCD 6桁表示
●表●オ ー ト オ フ：約20分●最大応答速度：右方向へのスライド…1.5m/s
左方向へのスライド…0.75m/s※右方向、左方向は各部の名称の図を正面にした場合●使用温度範囲：
0°C〜40°
C
C〜60°C●保管温度範囲：−10°
源：リチウム電池 CR2032×1●電●電 池 寿 命：通常使用状態で約2.5年●付
属品：・リチウム電池 CR2032×1（テスト用）
・マスタプレート

マスタプレートの検査成績表を準備してください。

②“P”の文字が表示されている状態で、“ON/OFF/P”スイッチを長押しすると、点滅位置が変わります。

入れ込む

端子※マスタプレート検査成績表に記載の寸法 値のプリセット設定
は、ご購入後初めて使用される場合・電池交換を行った場合のみ必要です。
※電源をオフにしてもプリセット値は記憶されますので、毎回設定する必要はありません。

①“ON/OFF/P”スイッチを押し、放すとLCD表示部の“P”の文字が点滅します。

ツメ

トラブルと対応⑦電池蓋 ………………… 開けて電池をセットします。

してはいけない内容『禁止事項』です。

■マスタプレート寸法値のプリセット設定① 測定面①測定面 …………… この面を被測定物に当てて測定します。

安全上のご注意

電池蓋

電池⑪マスタプレート⑧ZERO/INCスイッチ⑨ ON/OFF/Pスイッチ①電源をONにするには、“ON/OFF/P”スイッチを押してください。
②電源をOFFにするには、”ON/OFF/P”スイッチを長押し（2秒以上）してください。
③電源ONの状態から“HOLD”スイッチを押すと数値が固定されます。

●電池をセットする際に上から強く押し込むと、端子が折れて破損する場合があります。
●電池をセットする際は、右斜め方向から端子を外側に押すように、本体のツメの下へ入れ込んでください。

押し込む

⑦ 電池蓋

電池は必ずCR2032をお使いください。

●電池をセットする際は、端子の折れにご注意ください。

⑤ HOLDスイッチ④ 止めねじ■電源のON/OFF

す外ご注意●正しく安全にお使いいただくため、ご使用の前に本取扱説明書を必ず読み、記載の手順に従ってご使用ください。
お読みになった後は、お使いになる方がいつでも見ることができる場所に大切に保管してください。
●第三者に譲渡・貸与される場合も、この説明書を必ず添付してください。
●当商品に関するお問い合わせは、お買い求めの販売店もしくは当社にご連絡ください。
③ LCD表示部

使用方法①電池蓋を矢印の向きに開けます。
②電池を、必ず（＋）側を表にしてセットしてください。
③電池蓋を元通りに装着してください。

この度は「穴ピッチ用デジタルノギス」をお買上げいただきありがとうございます。
この商品は穴と穴の芯間、または端面から穴芯までの寸法を測定する精密測定工具です。

■各部の名称と機能電池のセット・交換③点滅している位置で、一度 “ON/OFF/P”スイッチを放し、再度短く押すと数値が変わります。
④ ②③ を繰り返し、マスタプレート同梱の検査成績表に記載されている寸法値を、LCD表示部に表示させてください。

※検査成績表には小数点以下第3位まで記載されています。第3位を四捨五入した数値をLCD表示部に表示させてください。

⑤点滅位置を“P”に戻し、“ON/OFF/P”スイッチを再度短く押してください。これで数値がプリセットされました。

■原点の設定※原点の設定は、毎回のご使用前に必ず行ってください。

付属のマスタプレートを使用し、通常測定用の原点を設定します。
①ジョウ先端の測定面と、マスタプレートをパーツクリーナーなどの洗浄剤で洗浄します。
②止めねじをゆるめ、電源を入れてください。

③LCD表示部に“P”が出ていない状態で、付属のマスタプレートを測定し（図 1、図 2）、“ON/OFF/P”スイッチを2回押してください。
（ 図 3）

Ｑ．正しい数値が表示されない。
Ａ．使用環境の湿度が高い可能性があります。本器を乾燥させ、湿度が低い環境下でご使用ください。

④マスタプレートの検査成績表の寸法値と、LCD表示部の数値が一致していることを確認してください。
これで原点が設定されました。計測を開始してください。

修理の際は（図2）●正常に作動しない場合や、不明な点がございましたらお買い上げの販売店または、当社までご連絡ください。

●お問い合わせや、ご連絡が無いまま直接当社に修理品などを送付されても処理、対応ができない場合がありますのでご了承ください。

使用後のお手入れ・保管方法●測定面・摺動面・外観などの汚れや油などを乾いたウエスで拭き取り、防錆処理を行う
防錆処理は防錆油を薄く塗るか、防錆袋などに入れてください。

●止めねじをゆるめた状態で保管する止めねじを締めて固定すると、温度変化による膨張などで無理な力がかかり、精度不良の原因となります。

●付属のケースに入れ、乾燥した冷暗所に保管する
直射日光や湿気は避け、管理者以外が触れない状態で保管してください。

●長期間使用しない場合は、電池を製品から取り外した状態で保管する
長期間使用しないで放置すると、電池の液漏れの原因となります。

（図1） マスタプレート●GDCP-200AAAB

GDCP-200TA GDCP-200AA
測定位置
測定位置

１ 測定

マスタプレート
（断面図）

（図 3）
●GDCP-200AA・200TA共通●GDCP-200TA

１ 測定

２ 短く2回押す

マスタプレート（断面図）〒955-0055 新潟県三条市塚野目5丁目3番14号(0256)33-5502(代) FAX(0256)33-5528

URL http://www.niigataseiki.co.jp

2401

I358-K1