User Manual

Version: WireXpert4500_Copper_IT_EN_202002© Copyright 2020 Softing Singapore Pte Ltd

Disclaimer of liability
The information contained in these instructions corresponds to the technical status at the time of printing of it and is passed on with thebest of our knowledge. The information in these instructions is in no event a basis for warranty claims or contractual agreements
concerning the described products, and may especially not be deemed as warranty concerning the quality and durability. We reserve theright to make any alterations or improvements to these instructions without prior notice. The actual design of products may deviate fromthe information contained in the instructions if technical alterations and product improvements so require.

It may not, in part or in its entirety, be reproduced, copied, or transferred into electronic media.

Softing Singapore Pte Ltd
73 Science Park Drive#02-12/13 Cintech I
Singapore Science Park 1
Singapore 118254
http://itnetworks.softing.com+ 65 6569 6019
+ 65 6899 1016
asia-support.itnetworks@softing.com
The latest version of this manual is available in the Softing download area at: http://itnetworks.softing.com.

Table of Contents

Table of Contents
Table of Contents........................................................................................................... 3
Introduction ................................................................................................... 6
1.1
1.2
1.3
1.4
1.5
1.6
1.7

About product ............................................................................................................................. 6
Safety precautions ...................................................................................................................... 6
Intended use ............................................................................................................................... 7
About this document .................................................................................................................. 7
Before you start .......................................................................................................................... 8
Scope of delivery......................................................................................................................... 8
System requirements .................................................................................................................. 9

WireXpert PC Software and Firmware .......................................................... 10
2.1
2.2

Install software ......................................................................................................................... 10
Upgrade firmware ..................................................................................................................... 10

Understanding WireXpert............................................................................. 12
3.1
3.2
3.3
3.4
3.5
3.6
3.7
3.8
3.9

WireXpert Overview ................................................................................................................. 12
Dual Control System (DCS)........................................................................................................ 13
WireXpert Adapters .................................................................................................................. 13
Cabling Certification and Testing .............................................................................................. 14
Memory..................................................................................................................................... 16
Battery Information .................................................................................................................. 17Power Supply ............................................................................................................................ 18
Dimension ................................................................................................................................. 18
Environmental Conditions ........................................................................................................ 18

WireXpert User Interface .............................................................................. 19
4.1
4.2

Touch Screen Layout ................................................................................................................. 19
The One Touch Access Button .................................................................................................. 20

Setting Reference ......................................................................................... 41
Configuring an AUTOTEST ............................................................................. 42
6.1
6.2
6.3

Setting up WireXpert ................................................................................................................ 42
Testing Guide for Permanent Link Testing ............................................................................... 43
Testing Guide for Channel Testing ............................................................................................ 44

Performing an AUTOTEST ............................................................................. 45
7.1
7.2
7.3
7.4

Understanding AUTOTEST ........................................................................................................ 45
NEXT and RL Fault Locating....................................................................................................... 47
Managing test result(s) ............................................................................................................. 48
Exporting test results into eXport PC Software ........................................................................ 50

Declarations ................................................................................................. 53
Appendix ...................................................................................................... 56
Glossary........................................................................................................ 58© Copyright 2020 Softing IT Networks

3

Table of Contents

Related documents....................................................................................... 61
Technical Support ......................................................................................... 62

4© Copyright 2020 Softing IT Networks

Table of Contents

This page is left intentionally blank© Copyright 2020 Softing IT Networks

5Introduction

Introduction
1.1 About productSofting's WireXpert, with its unparalleled 2,500 MHz measurement range, is the first cable certifier withcapability to certify the highest performance cabling systems in enterprise networks and data centers.
Cable installers make significant gain in productivity with WireXpert‘s industry leading test speed andease of use. With certification testing up to Class FA and CAT8 copper cabling, as well as MPO, SM, MM
and MMEF fiber optic cabling, WireXpert is ready for 40G and beyond.
Starting from v7.4, WireXpert is available in four variants- WireXpert 500 copper (up to 500MHz) only,
WireXpert 500 fiber only, WireXpert 500-plus; copper + fiber and WireXpert 4500- to provide more
flexibility to suit your certification requirements. The WireXpert 500 series provides an affordable
solution that can grow with your cable certification needs. Upgrade the copper only or fiber only
variants to the combined WireXpert 500-plus or WireXpert 4500 for the full 2,500MHz capability.
With v8.0, cloud-based data management features have been integrated into WireXpert and eXport PC
Software. Users will be able to connect unlimited numbers of WireXpert and transfer test data to andfrom their private network using Wi-Fi or mobile hotspot. The cloud feature is available in all variants of
WireXpert.

1.2 Safety precautions
Read this manual before startingFor damages due to improper connection, implementation or operation Softing refuses
any liability according to our existing warranty obligations.

Note
This symbol is used to call attention to notable information that should be followed
during installation, use, or servicing of this device.

Hint
This symbol is used when providing you with helpful user hints.

CAUTION
Selection of option may cause all or partial of saved data and/or settings in the device
to be erased or restored to non-reversible original factory state. Backing up of saved
result(s) is recommended before executing option.

6© Copyright 2019 Softing IT Networks

Introduction

CAUTIONCAUTION indicates a potentially hazardous situation which, if not avoided, may resultin minor or moderate injury.

WARNINGWARNING indicates a potentially hazardous situation which, if not avoided, could result indeath or serious injury

DANGERDANGER indicates an imminently hazardous situation which, if not avoided, will result indeath or serious injury. This signal word is to be limited to the most extreme situations.

1.3 Intended use
WireXpert series has been designed for use in factory, process and building control. The unit must not
be used in explosion hazard areas. The permissible ambient conditions given in the Technical Data must
be complied with.
The faultless and safe operation of the product requires proper transport, proper storage andinstallation, and expert operation and maintenance in accordance with the manual.

1.4 About this document
Read this manual before startingFor damages due to improper connection, implementation or operation Softing refuses
any liability according to our existing warranty obligations.

1.4.1 Conventions used
The following conventions are used throughout Softing customer documentation:
Keys, buttons, menu items, commands and otherelements involving user interaction are set in bold
font and menu sequences are separated by an
arrow

Open Start  Control Panel  Programs

Buttons from the user interface are enclosed inbrackets and set to bold typeface

Press [Start] to start the application

Coding samples, file extracts and screen output
are set in Courier font type

MaxDlsapAddressSupported=23© Copyright 2020 Softing IT Networks

7

Introduction

Device description files are located in
C:\<productname>\delivery\software\Device
Description files

Filenames and directories are written in italic

1.5 Before you start
Check that the latest eXport PC software and firmware is installed in the workstation and WireXpert
respectively to ensure the latest features are available.
Key differences between WX4500 and WX500 series
Features

WX4500-FA

WX500-PLUS

WX500-CU

WX500-FIBER

Frequency ofmeasurement
Accuracy Specification

2500 MHz

500 MHz

500 MHz

N/A

TIA Level 2G
ISO/IEC Level VI
Yes
Yes
Yes

TIA Level IIIe
ISO/IEC Level
IIIe
No
No
Yes

N/A

Fiber Testing option
Class FA/CAT 8 options
Patch Cord Test adapters

TIA Level IIIe
ISO/IEV Level
IIIe
Yes
No
Yes

Yes
No
No

Softing recommends WireXpert to be calibrated annually. Starting from v7.4, WireXpert will remind theuser to calibrate the units if the last calibration was more than one year ago. This notification can beclosed by clicking the [OK] button, but will be displayed on every boot up until the units are calibrated.
Calibration can only be done by the service center. Please contact your local vendor for more
information.

1.6 Scope of delivery
WireXpert WX-4500-FA or WX-500-CU comprises the following parts:

Soft or Hard carrier case
(WX_AC_SOFTCASE) or(WX_AC_HARDCASE)

8

WireXpert WX4500 or
WireXpert WX500(LOCAL and REMOTE)

Calibration Certificate& USB Flash Drive© Copyright 2019 Softing IT Networks

Introduction

Power Adapters Input: 100-240VAC/1.2A
Output: 12VDC/3.0A(WX_AC_PWRAD)

Talk-set headphones(WX_AC_TALKSET1)LCD Protective Cover(WX_LCD_COVER)

CAT 6A Channel
adapters(WX_AD_6ACH2)

Permanent Link
adapters & cords(WX_AD_6ALKIT2)

Straps

1.7 System requirements
Hardware

PC

Operating system
Windows 7, 8.x or 10 (32 bit or 64 bit)Intel Core 2 Duo, 2GHz

1 GB of RAM



200 MB of free space of installation



Microsoft .NET framework© Copyright 2020 Softing IT Networks

9WireXpert PC Software and Firmware

WireXpert PC Software and Firmware
2.1 Install software
1

Download the latest eXport PC software from https://itnetworks.softing.com/

2

Run eXport setup v8.x.x64.exe

3

Follow through the instructions on the screen and click [Next] to proceed.

4

Click [Install] to begin installation.

5

Click [Finish] to complete installation.

6

If this is an upgrade from an earlier version from 6.x and above, click [Yes] to proceed.

7

Follow the instructions to complete installation.
User Manuals
For more information on installation and using eXport PC software, please refer to“Installation Guide for eXport PC software” and “User Manual for eXport PC software”.
Note
Softing IT Networks has ceased support for ReportXpert v5.x and earlier. Please contact
asia-support.itnetworks@softing.com for more information.

2.2 Upgrade firmware

10

1

Download and install the latest eXport PC software.

2

Connect an USB flash drive to the workstation.

3

Run eXport PC software.

4

Go to Tools  Update Device Firmware

5

Click [OK] and select USB drive from “Export to USB” window.

6

Click [Export] and [OK] to proceed.

7

Please wait while exporting takes place. This process may take a while.

8

Remove USB flash drive from workstation and connect to WireXpert.

9

Select [Upgrade Firmware] from prompt and click [OK] button to continue.
Note: Both LOCAL and REMOTE units must be upgraded.
© Copyright 2019 Softing IT Networks

WireXpert PC Software and Firmware

10 If prompt did not appear, check that USB icon is present on the status bar, and press the [SETUP]
button  System Settings  Settings 2 Storage USB.
11 Please wait while upgrading takes place. This process may take a while.
12 Upgrade process is complete.
CAUTION
Saved test results and settings may be erased during upgrading. You are recommended
to save all test results before upgrading the firmware.

© Copyright 2020 Softing IT Networks

11Understanding WireXpert

Understanding WireXpert
3.1 WireXpert Overview
Power ON/OFF button
Touch sensitive color LCD screen
USB device port
Audio port
USB host connector for flash drive
RJ-45 Ethernet port for remote control
Power supply port (12V)
One-Touch access buttons
Brightness control button
Navigation scroll buttons
Context sensitive Help button
Probe interface connector
Battery compartment

Part

Functionality

Power ON/OFF Button

To turn on the device, press the power button for 5 seconds.
To turn off the device, press the power button, and select OK on the
power off dialog box. Alternately, press and hold the power button for
10 seconds (not recommended). The LED in the power button turns
green if the unit is turned on.
The LED on the power button turns red when the device is charging instand-by mode, orange if it’s turned on.
Performs test executions and viewing of results.

Touch Sensitive LCD
screenUSB device port
Audio port
USB Host port
RJ-45 Ethernet port
Power Supply port
One-touch access buttons
Brightness Control
Navigation scroll buttons
Context sensitive help
12

Enables remote connection to the PC for remote control update andexporting saved results.
Enables vocal communication between LOCAL and REMOTE units whenconnected to talk set headphones (WX_AC_TALKSET).
Enables results to be exported and firmware upgrade via USB flash
drive.
Enables loading of custom limits and labels for list-based testing.
Enables communication between two sets of WireXpert for Alien
Crosstalk testing. Currently not available.
Performs charging of device when connected to power adapter shippedwith WireXpert. (WX_AC_PWRAD)
Main functional buttons AUTOTEST, SETUP, DATA and TOOLS for
WireXpert. Enables quick AUTOTEST and results viewing in a single click.
Adjusts the brightness level of the LCD display.
Performs navigation in long lists.
Provides simple explanation relevant to the functionality on the current
screen (available in future version).
© Copyright 2019 Softing IT Networks

Understanding WireXpert

Part

Functionality

Probe interface
connector

When connected with different probe adapters (e.g. CAT6A channel or
Permanent Link), WireXpert will enable specified certification test to be
conducted.
Battery storage compartment. Unscrew to replace battery(WX_AC_BAT1).

Battery compartment

3.2 Dual Control System (DCS)
WireXpert consists of two identical devices, LOCAL and REMOTE, for dual-end testing and certification.
Both devices utilize easy to use LCD touch-screen technology and graphical user interface. The DCS
ensures the same graphical data on the LOCAL units are received on the REMOTE unit real-time,
improving efficiency and productivity significantly as test data and settings can be viewed from REMOTE
and LOCAL side.
In the event of a wire map failure, the DCS will enable the operator on the far-end to visually identify
which pair of wires is causing the error, instead of just a LED indication on traditional testers.
2m

30m

2m

10 m

2m

Traditional tester with single display screen, usually
on LOCAL unit only

2m

30m

2m

10 m

2m

WireXpert with LCD screens display on both LOCAL
and REMOTE units

3.3 WireXpert Adapters
All test adapters are designed to fit onto every WireXpert units. The device automatically detects
adapters when connected and prompts to change in the matching interface depending on the type of
adapter. Press the [SETUP] button  Quick Setup to change the test limit using generic configurations.

3.3.1 Permanent Link (PL) and Channel (Ch) Adapters
Every standard WireXpert kit includes adapters for testing copper cablings in Permanent Link andChannel configurations.
Channel adapters connect to the patch cords at the wall plates and telecommunication panels.
WireXpert software compensates for the adapter’s transmission characteristics to ensure test results’
accuracy.
Permanent Link adapters connect to the wall plates and telecommunication panel jacks. WireXpert
software compensates for the adapter’s transmission characteristics to ensure test results will not beaffected.
WireXpert’s Permanent Link adapters’ high precision interface reduces the degrading issue between the
RJ-45 jack and link cord, providing highly accurate and repeatable measurements, and cost-efficientmethod of replacing the link cords instead of the expensive link adapters.

3.3.2 Adapter Configuration Warnings
WireXpert will display a warning a potential conflict is detected between the adapter and the test
settings. It is important to use the appropriate adapters as accuracy of measurements will be affected.

© Copyright 2020 Softing IT Networks

13Understanding WireXpert

3.3.3 Adapter Insertion Counts
Every WireXpert adapter can keep track of the number of tests performed. This data ensures up to date
information on the adapter’s lifespan. WireXpert will indicate a warning when the adapter insertion
counts have expired the specified insertion counts lifespan. This can be checked by pressing the [SETUP]
button  System Settings  System Settings 2  Device Info  Probe Test Count

3.4 Cabling Certification and Testing
Certification testing with WireXpert is highly intuitive and simple. The AUTOTEST function performs aseries of measurements between the LOCAL and REMOTE devices and analyses the results to determine
if the cabling run passes or fails the selected standards.

3.4.1 Copper Cable Certification Testing
Test at both ends of structured twisted pair cables for the worst-case performance is required inaccordance to the ISO/TIA standards. All certification testing requires a two-part test consisting of amain and remote unit, both bearing similar test capabilities. The test would be conducted with one end
from the telecommunication closet, and the other at the outlet.
There are two primary considerations regarding the test method:••

Whether the user patch cords are included in the cabling run during the test.


If so, channel test configuration is required.



If not, link test configuration is required.

Which standard – TIA or ISO should be used (Category or Class)3.4.2 Permanent Link Configuration Tests
The Permanent Link configuration is generally used in facilities still under construction and does notinclude user patch cords. As the link configuration does not include the two additional connections of
the patch cords, performance standards are more stringent. Special link adapters are used on both
LOCAL and REMOTE units for links that are under test.

3.4.3 Channel Configuration Tests
User patch cords are included for the Channel configuration. The pass/fail limits for this configuration
allows performance degradation inherent in the two additional patch cords included in themeasurements, thus are less stringent. Special channel adapters are used on both LOCAL and REMOTE
units for channels that are under test.

14© Copyright 2019 Softing IT Networks

Understanding WireXpert

3.4.4 End-to-End (E2E) Link Configuration Tests
1 Segment(s); 2 Connections (V)2 Segment(s); 3 Connections

3 Segment(s); 4 Conn.
(Coupler as 1 Conn.)

1 Connection

1 Connection

3 Segment(s); 4 Conn.
(Coupler as 2 Conn.)

2 Connections

2 Connections

3 Segments, 4 Connections

4 Segments, 5 Connections

5 Segments, 6 Connections

Starting from v7.3, WireXpert has included the E2E test limits to certify links that are more suitable forthe industrial environment. E2E links are applicable when application is required to connect directly tomachines or machines to network devices. The Industrial Ethernet test kit is required to perform a E2Elink test. The Industrial Ethernet test kit consist of a pair of End-to-End test adaptors and test cords of
TERA-TERA, TERA-RJ45, TERA-M12 D-coded or TERA-M12 X-coded variation depending on theapplication requirement. Press the [SETUP] button  Test Settings  Test Limit  End to End toselect the required E2E test limit.
The Industrial Ethernet E2E test kit is not part of the standard WireXpert kit. Please contact your local
vendor for more information.
User Guide
Refer “Application Notes - E2E Link Test” for more information.

© Copyright 2020 Softing IT Networks

15

Understanding WireXpert

3.4.5 Modular Plug Terminated Link (MPTL)
Fixed Installed Cabling

Permanent Link

Horizontal Cabling

Test Cord
INTERCONNECT

MODULAR PLUGTERMINATION

Permanent Link Adapter

Patch Cord Adapter

Horizontal CablingCPwith Consolidation Point
INTERCONNECT

MODULAR PLUGTERMINATION

Correct measurement of MPTL link

Starting from v7.4, WireXpert has included the MPTL test limits to certify links that features a socket on
the patch panel, and a plug on the opposite side of the link. This is similar to the E2E link, however, isused when IP-capable devices are permanently installed, e.g, LED lights, building automation elements
or surveillance cameras. The installation cable is plugged directly into the termination deice without theaddition of a socket plus patch cable. A Permanent Link and a Patch Cord test adapter is required toperform the MPTL test. Press the [SETUP] button  Test Settings  Test Limit  TIA to select therequired MPTL test limit.
The Patch Cord adapter is not part of the standard WireXpert kit. Please contact your local vendor for
more information.
User Guide
Refer “Application Notes - MPTL” for more information.

3.5 Memory
1 Gigabyte internal flash memory.
External USB Flash Drive included allows for flexible data management and storage.
WireXpert only supports FAT/FAT32/exFAT file system.

3.5.1 Memory Requirements
The amount of memory that each test result takes up depends on the options that is configured before
starting the test. If plots are saved with each test, file size of test results will be generally larger than one
that has plots option omitted.
WireXpert allocates approximately 1 GB of internal flash memory for test result storage.

16© Copyright 2019 Softing IT Networks

Understanding WireXpert

3.6 Battery Information
WireXpert units are powered by Lithium Ion rechargeable batteries. These batteries contain circuitrythat reports their state of charge to WireXpert. The units can be powered by external AC/DC power
adapters.
A fully charged battery can operate WireXpert for five to eight hours before requiring recharging.
Reducing the screen brightness and enabling the sleep function allows WireXpert to run longer on acharge.

WARNING
Do not set reference or operate WireXpert while the battery is hot immediately after
charging.
Make sure that the WireXpert is always operated within the allowed temperature range
from 0-40°C. If a significant increase or decrease of the temperature is discovered, repeat
the set reference procedure to remain the instrument’s accuracy.
CAUTIONUse only the supplied power supplies with the WireXpert kit. Using otherpowers may damage the tester and void its warranty.
3.6.1 Battery Safety•

To avoid the risk of fire, burns or damage to your battery pack, do not allow metal objects to touch thebattery contacts.

•

The battery pack is suitable for use only with compatible WireXpert devices.

•

Do not disassemble the battery pack. There are no user serviceable parts inside. Do not dispose thebattery pack in fire or water.

•

Handle a damaged or leaking battery with extreme care. If you come into contact with the electrolyte,
wash the exposed area with soap and water. If the electrolyte contacts the eye, flush the eye withwater for 15 minutes and seek medical attention.

•

Do not expose the battery pack to high storage temperatures (above 55°C).

•

When discarding a battery pack, contact your local waste disposal provider regarding local restrictions
on the disposal or recycling of Lithium Ion batteries.

•

To obtain a replacement battery (part number WX_AC_BAT1), contact your local dealer.

•

Do not charge the batteries pack if the ambient temperature is above 40°C.

3.6.2 Replacing Battery
The batteries are located and secured at the bottom of the WireXpert units for safety reasons. To
replace the batteries;
1. Switch off WireXpert and disconnect external power supplies.
2. Unscrew the battery compartment and remove the cover.
© Copyright 2020 Softing IT Networks

17Understanding WireXpert

3. Remove and replace a new battery into the compartment.
4. Close, fasten and secure the screw onto the compartment cover.

3.7 Power Supply•

Removable/rechargeable Li-Ion batteries provide more than 8 hours of continuous operation.

•

AC power: 100~240V AC plugs directly into the battery. Alternatively, the batteries can be recharged using
an external charging kit that can be ordered separately (WX_AC_CHARGER).

3.8 Dimension•

Size: 223mmH x 124mmW x 74mmD•

Weight: approx. 1.0 kg (approx. 2.2 lbs.)

3.9 Environmental Conditions•

Operating temperature: 0°C to 40°C•

Storage Temperature: 20°C to 60°C•

Relative humidity: 10% to 80%

H

W

18

D

© Copyright 2019 Softing IT Networks

WireXpert User Interface

WireXpert User Interface
Starting from v7.3, users are required to read and agree to the “End User License Agreement (EULA)” on
the first launch or after a firmware upgrade to use WireXpert. The EULA can be found in [TOOLS] 
About Us  Info after agreeing the first time.

4.1 Touch Screen Layout
The Graphical User Interface (GUI) in version 7.0’s firmware has been updated with a more responsive
system and quick-access menus.
WireXpert boots up to the SETUP screen. It is categorized into 5 groups:
1. Status bar
2. Quick Setup group

3. Test Settings group

4. Project Settings group

5. System Settings group

1. The Status bar displays the current date, Wi-Fi connectivity status, USB connectivity status, talk set
connectivity status and battery level. Tap icons to change or view setting.
2. The Quick Setup group provides quick selection with predefined configurations.
3. The Test Settings group provides results-oriented configurations necessary to perform an AUTOTEST.
4. The Project Settings group provides non-results-oriented configurations before performing an
AUTOTEST.
5. The System Settings group settings provides device, time, localization and device related
configurations.

© Copyright 2020 Softing IT Networks

19

WireXpert User Interface

4.2 The One Touch Access Button

The fundamental philosophy behind the WireXpert User Interface is simplicity in its ease of use. The
main functions of the One-Touch access buttons as follows:

4.2.1 The [AUTOTEST] button
The [AUTOTEST] button will perform an immediate certification test on the lastconfigured settings. If no settings were configured, default settings will be used.
Test results will be generated automatically after the test is completed.
You will receive any of the following 4 results after performing AUTOTEST:Green “PASS” – Good test result in accordance to pre-defined settings.
Green “PASS*” – “PASS” result with one or more test parameters that have margin in the rangeof uncertainty of the tester.
Red “FAIL” – Unacceptable results with severe disturbance on one or more test parameters.
Red “FAIL*” – “FAIL” result with one or more test parameters that have negative margin in therange of uncertainty of the tester.
Note
Marginal Pass/Fail is enabled by default. Press the [SETUP] button  Test Settings 
Test Options  Lab  Marginal Pass/Fail to change the presentation of the test result
to clear “PASS” or “FAIL” results. Uncertainty of the tester is then not taken into
consideration. Contact Softing for more information.
You will be given the following option after performing an AUTOTEST:•[Retest] can be conducted on a PASS or FAIL result (if list-based testing labelling scheme is selected).

•[View] details will display additional test parameters and plots.

•[Save] test results to device

An AUTOTEST will fail in the event of missing connection between the LOCAL and REMOTE units, wrong
settings configured or if setting reference are required. WireXpert will perform an initial wire map test
before testing other parameters. Disable wire map test at Test Settings  Test Options  Lab  Wire
map to bypass this test.

20© Copyright 2019 Softing IT Networks

WireXpert User Interface

AUTOTEST results[View] detail results

Plots

Two worst tested parameters will be displayed on the result summary screen. Press the [View] button
to view detailed results.
The type of results will differ depending on the Test Limits and type of cable.
Test results on the screenshots are for illustration purposes only and not to be taken as reference values.
4.2.1.1

The Tone Generator mode
Starting from v8.0, unless when performing a Single-Ended cable test, WireXpert will automaticallyswitch to Tone Generator mode when only one end is connected and the [AUTOTEST] button is pressed.
WireXpert will be able to perform audible cable detection when paired with a Softing CT15 Tone
Generator.
When faced with multiple unlabeled open-ended cables, connect one end of the cable to WireXpert.
Point the CT15 Tone Generator to the other end of the cables. CT15 will produce a different tone when
the cable that is connected to WireXpert is detected.
The Tone Generator mode can also be triggered by pressing the [TOOLS] button  Tone Generator.

4.2.2 The [SETUP] button
The [SETUP] button provides setting options necessary to conduct an AUTOTEST and configure
the device.
These options include –4.2.2.1

Quick Setup
Quick Settings provides quick selection using predefined TIA (CAT 5e/6/6A) and ISO (Class D/E/EA/F/FA)
configurations for UTP/FTP cables. WireXpert will detect if Channel or Permanent Link adapter is usedand changes its test limits automatically.

© Copyright 2020 Softing IT Networks

21

WireXpert User Interface

4.2.2.2

Test Settings
Test Settings provides results-oriented configurations necessary to perform an AUTOTEST including; -4.2.2.2.1 Test Limit
Choose from a list of standards to determine the performance criteria in a given category or class. Threeof the most commonly used limits will be listed in the “Recent Limit” list.
Press the [SETUP] button  Test Settings  Test Limits to choose from a list of commonly used limits.
Three recently used limits will be on top of the list.
Test Limits update
Starting from v7.3, the following limits has been added into WireXpert••

TIA – Cat 8 ChannelTIAo22-23 AWGo

24 AWGo

26 AWGo

Cat 5e MPTL (Modular Plug Terminated Link)o

Cat 6A MPTLo

ISO – Five Connections Class D E2E Linko

ISO – Four Connections Class D E2E LinkoISO - Three Connections Class D E2E Linko

ISO – Two Connections Class D E2E Link

o

•

22

Cat 6 MPTL

End-to-End (E2E)oISO - Five Connections Class E E2E Linko

ISO – Four Connections Class E E2E Linko

ISO – Three Connections Class E E2E Linko

ISO – Two Connections Class E E2E Link

© Copyright 2019 Softing IT Networks

WireXpert User Interface•

Application Standards
o1000 Base-TX

o

10 Base-To

100 Base-TX•

Cenelec EN 50173 Class FA Channel•

Taiwan EL-3600-9 Standards•

China GB_T 50312-2016 Standards

Standard test limit list

4.2.2.2.2 Cable
Choose from a list of cable manufacturer for more specific test parameters. If unsure of manufacturer,choose [Generic UTP] or [Generic Shielded] or [Customized Cable] to create custom cable.
Press the [SETUP] button  Test Settings  Cable Manufacturer to choose from a list of cablemanufacturers. If unsure of manufacturer, choose [Generic UTP] or [Generic Shielded], or [Customized
Cable] to create custom cable.

Generic UTP cable list
© Copyright 2020 Softing IT Networks

Generic Shielded cable list
23

WireXpert User Interface

Customized Cable
Click [Add] to add or Manage  [Delete] to remove customized cable(s) from the customized cable list.

Customized cable list

Adding customized cables

When creating customized cable, determining the cable name, number of pairs, construction type,
performance grade(S/FTP) and cable’s NVP is required.
WireXpert introduces the [Manage] button on version 7.0 firmware.
[Manage] enables multiple saved items such as sites, operators, customized
cables, customized connectors and results to be selected and deleted
simultaneously.
4.2.2.2.3 NVP
Determine the Nominal Velocity of Propagation (NVP) value of the cable. The NVP factor is required todetermine the accurate length of the cable run to be measured. Based on this constant value, the
WireXpert performs its TDR calculations. Please see the cable’s data sheet for the specific value of thecable type in use.
If the specific NVP value is unknown, press the [TOOLS] button  Learn NVP to determine the value
using the predefined cable run of at least 30 meters or 100 feet.
4.2.2.2.4 Connector
Press the [SETUP] button  Test Settings  Connector Manufacturer to choose from a list ofconnector manufacturers. If unsure of manufacturer, choose [Generic Unshielded] or [Generic
Shielded], or [Customized Connector] to create custom connector.
Click [Add] to add or Manage  [Delete] to remove customized connector(s) from the customized
connector list.

24© Copyright 2019 Softing IT Networks

WireXpert User Interface

Add customized connector

Manage customized
connector

4.2.2.2.5 Advanced Test Options
Advanced Test Options determines how the test will be conducted and results will be displayed.
General© Copyright 2020 Softing IT Networks•

Locator – If enabled, WireXpert will display NEXT and Return Loss (RL)
fault information in an AUTOTEST result. Enabled by default.

•

Cable Pairing Type – Choose T568A or T568B copper cable wiring
standard. “B” is selected by default. Option will be disabled when a
TERA Channel or TERA Link adapter is connected to WireXpert.

•

AUTOTEST On Connection – If enabled, WireXpert will perform an
AUTOTEST automatically when LOCAL is connected to the REMOTE
unit. Disabled by default.

•

Direct Attach – If enabled, WireXpert will perform Direct Attach
copper installation test. Attach a Permanent Link adapter to the
LOCAL and a Patch Cord adapter to the REMOTE unit to perform thistest. Connect the LOCAL unit to patch panel and REMOTE to fieldterminated end. Disabled by default.

25

WireXpert User Interface

Patch Panel
Field
Termination

Permanent Link
Adapter

SETUP

DATA

Patch Cord
Adapter

TOOLS

SETUP

DATA

TOOLS

AUTOTESTAUTOTEST

Ȝ

26

Ȝ

•

AC Wire map – If enabled, WireXpert is able to measure test runs with
Power over Ethernet (PoE) midspan devices in between. The
WireXpert supports cable runs with IEEE 802.3 af and 802.3 atinjectors. Disabled by default. Option will be disabled when Shield
Test option is enabled.

•

Crossover – If enabled, depending on the limits selected, WireXpert isable to test one or more of the follow crossover cable. Disabled bydefault.

Straight-throughcable

Crossover at pairs
12, 36

Crossover at pairs
12, 36, 54, 78

Crossover at pairs
12, 36 and reversed
at pairs 54, 78

Crossover at pairs
12, 36 for 2-pair
cable

•

DCRU – If enabled, WireXpert will perform additional DC resistanceunbalance test measurements between and within each pair inaccordance with TIA and IEC specifications, on top of the AUTOTEST
parameters required in a Channel or Permanent Link. Disabled bydefault. Option is only available with CAT6A+ adapters.

•

Shield Test – If enabled, WireXpert will perform shield integrity test to
identify if a shield is present and connected between both ends on
the actual cable under test. Disabled by default. Option is only
© Copyright 2019 Softing IT Networks

WireXpert User Interface

available with CAT6A+ adapters. Option will be disabled when AC
Wiremap option is enabled.

Note
Crossover can only be selected by selecting a test limit from Test Settings  Test Limit Application Standards. Disabled by default.
Crossover wiremap selections will only be available for the applicable limits as follows:
Straightthrough cable

Crossover atpairs 12, 36

Crossover atpairs 12, 36,
54,78

Crossover atpairs 12, 36
and reversed
at pairs 54, 78

Crossover atpairs 12, 36
for 2-paircable

1000BASE-T

Yes

No

No

Yes

No

1000BASE-TX

Yes

No

Yes

No

No

100BASE-TX

Yes

Yes

Yes

Yes

Yes

10BASE-T

Yes

Yes

Yes

Yes

Yes

User Guide
Refer “User Manual – CAT 6A+ Adapters” for more information.
Lab
Laboratory use only. Password is required to access this section. Please contact your local WireXpert
Vendor for more information.

© Copyright 2020 Softing IT Networks•

Wiremap Test – If enabled, WireXpert will not continue
with the AUTOTEST process if a wire map FAIL is
detected. Enabled by default.

•

Marginal Pass/Fail –If enabled, WireXpert will display
marginal pass (PASS*) and marginal fail (FAIL*) results.
A marginal test result is obtained when its margin issmaller than the accuracy specification for the test
parameter. Enabled by default.

•

Save Phase Data – If enabled, WireXpert will generate
information for phase data plot generation. Export
saved results to CSV format using eXport software toobtain phase data. Impedance plot will be displayed on
eXport. Disabled by default.

•

3/4 dB Rule – If enabled, WireXpert will ignore Return
Loss (RL) measurements if the Insertion Loss (IL) is<3dB and NEXT measurements if the IL is <4dB.
Disabled by default.

•

Single Ended Test – If enabled, WireXpert will be able
to perform a cable test to obtain only Length & Delay,
RL, NEXT and PS-NEXT results without a REMOTE unit.
Single Ended Test can only be performed using a27

WireXpert User Interface

Custom Limit. Refer to “User Guide – Custom Copper
Limit” for more information. Disabled by default.
4.2.2.3

Project Settings
Project Settings provides non-results-oriented configurations before performing an AUTOTEST.

4.2.2.3.1 Site
To add a site,
1. Press the [SETUP] button  Project Settings  Site to begin adding a new site.
2. Click the [Add] button to add a new site.
3. Enter a Site Name. Site Name is a required field. Test results and label source will be saved under
the created Site.
4. Site Address and Site Notes are optional fields, but they provide additional information of the site.
5. Press [OK] to save site and proceed.
By default, WireXpert creates an “UNSPECIFIED” site for default saving if new sites are not created.

To delete a saved site,
1.

Press the [Manage] button.

2.

Select site(s) and press the [Delete] to delete site(s).
WireXpert introduces the [Manage] button on version 7.0 firmware.
[Manage] enables multiple saved items such as sites, operators, customized
cables, customized connectors and results to be selected and deleted
simultaneously.

4.2.2.3.2 Operator
To add an operator,
1. Press the [SETUP] button  Project Settings  Operator to begin adding an operator.
2. Click the [Add] button to add a new operator.
28© Copyright 2019 Softing IT Networks

WireXpert User Interface

3. Enter Operator Name.
4. Press [OK] to save site and proceed.
By default, WireXpert creates an “UNSPECIFIED” operator for default saving if new operators are notcreated.

To delete a saved operator,
1. Press the [Manage] button.
2. Select operator(s) and press the [Delete] to delete operator(s).
4.2.2.4

Label Source
Press the [SETUP] button  Test Settings  Label Source to choose from a list of schemes.

None
Manual input of label names is required for every saving of result.
Simple Label
Provides simple numeric increments to a prefix. To configure the simple label scheme:© Copyright 2020 Softing IT Networks

29

WireXpert User Interface

1.

Label Name - Enter desired Prefix (e.g. Office, Company X) for every label.

2.

Start Value - Set a numeric start value (e.g. 1, 33, 245) for incremental counts.

TIA-606-A Class 1 (Single Room Horizontal Link)
TIA-606-A Class 1 standard labelling scheme mainly used to support labelling for a building or premisesthat is served in a single equipment room or a horizontal cross-connect.
The default “1A-A1” generally refers to “Floor + Tel Room – Panel + Port”.

TIA-606-A Class 2 (Single Building Horizontal/Single Building Backbone Cable)
TIA-606-A Class 2 standard labelling scheme mainly used to support labelling of infrastructure with one
or more telecommunication spaces in a single building.
The default “1A-A1” for Horizontal Link generally refers to “(1) Floor + (A) Tel Room – (A) Panel + (1)Port”.
The default “1A/1B-” for Backbone Cable generally refers to “(1) Tel Room 1 Floor + (A) Tel Room 1
Name / (1) Tel Room 1 Floor + (B)Tel Room 2 Name” followed by - “Backbone Cable . Cable”
30© Copyright 2019 Softing IT Networks

WireXpert User Interface

TIA-606-A Class 3 (Campus Backbone Cable)
TIA-606-A Class3 standard labelling scheme mainly used to support labelling for multiple buildings in asingle site.
The default “A-1A/A-1A-” generally refers to “(A)Building 1 Name - (1) Tel Room 1 Floor + (A) Tel Room 1
Name / (A) Building 2 Name – (1) Tel Room 2 Floor + (A) Tel Room 2 Name” followed by - “Backbone
Cable. Cable”

List Based Testing
List based testing allows creation of label list in the eXport software on PC and then bring the list to
WireXpert. It further allows easy selection of labels from the list to help technician select the cables tobe tested quickly. This testing method is carefully optimized for typical test work-flow, and it
significantly improves productivity.
1. Insert USB flash drive to USB port on WireXpert.
2. Device will automatically detect USB flash drive.

© Copyright 2020 Softing IT Networks

31

WireXpert User Interface

3. Select [Label List].
4. Select the custom label list(s) to be copied from the USB flash drive.
5. Select [Copy label files] to proceed.
6. Click [OK].

7. Press the [SETUP] button  Project Settings  Label Source  List Based Testing
8. Select label source file and click [OK] to continue.
User Manuals
Refer “User Manual – eXport” for more information on generating list-basedlabels.
9. Press the [AUTOTEST] button to display the loaded cable list.
10. Select label from list to begin AUTOTEST.
11. WireXpert will automatically save a PASS result and return to the list or allows you to save or
conduct a re-test for a FAIL result manually.
12. Select a tested label to view the results. A green result indicates a PASS and a red result indicates a
FAIL test. A re-test can be conducted by pressing [Retest].
13. Press the [AUTOTEST] button to return to the list.
Free Form
Free Form provides simple numeric increments to 2 or more prefix.
The default start “A-1” and end “B-5” labels will generate labels “A1 to A5 and B1 to B5”.

32

© Copyright 2019 Softing IT Networks

WireXpert User Interface

4.2.2.4.1 AutoSave

Enable option for WireXpert to automatically save every PASS result only. Rectify or manually save FAIL
results.
4.2.2.5

System Settings

4.2.2.5.1 Length Unit
Press the [SETUP] button  System Settings  Settings 1  Device Settings  Length Unit to selectthe preferred length unit to be displayed. Feet (FT) and Meters (M) are available.
4.2.2.5.2 Plot Y-axis Direction
Press the [SETUP] button  System Settings  Settings 1  Device Settings Plot Y-axis Direction to
select the how the plot’s Y-axis will be display. Normal and Inverted Y-axis are available.
4.2.2.5.3 Device Type
Press the [SETUP] button  System Settings  Settings 1  Device Settings  Device Type to switch
a LOCAL unit to REMOTE, or vice versa.
4.2.2.5.4 Sleep and Auto Power-off Interval
For Sleep Interval,
Press the [SETUP]button  System Settings  Settings 1  Sleep Interval to select the idle duration
before WireXpert enters the sleep (power saving) mode.
For Auto Power-off Interval,
Press the [SETUP] button  System Settings  Settings 1  Auto Power-Off Interval to select the idle
duration before WireXpert performs an automatic shutdown.

© Copyright 2020 Softing IT Networks

33

WireXpert User Interface

4.2.2.5.5 Audio Settings
Press the [SETUP] button System Settings  Settings 1  Audio Settings to configure various audio
options for WireXpert.
•

Touch Click – Enable option to trigger clicking sound options on screen touch.

•Speaker & Tone – Enable option to trigger system sound options, e.g., boot up.

•

Voice Prompt – Enable option to trigger system vocal notifications on action.

•

Volume Control – Adjust for audio loudness.

4.2.2.5.6 Date and Time
Press the [SETUP] button  System Settings  Settings 2 Date & Time to configure date, time anddisplay format.

34© Copyright 2019 Softing IT Networks

WireXpert User Interface

Date settings

Time settings

4.2.2.5.7 Language
Press the [SETUP] button  System Settings  Settings 2  Language to select preferred display
language.
4.2.2.5.8 Storage
Press the [SETUP] button  System Settings  Settings 2  Storage to view available storage space
on WireXpert and USB flash drive if inserted.
Read USB flash drive by clicking the USB icon if prompt did not appear or closed after inserting USB flash
drive.

Read USB flash drive

Select Upgrade Firmware to upgrade current firmware version. Please export all results before
performing an upgrade to prevent loss of data.
Select Test Results to export Sites to an USB drive. This may take a while depending on the number oftest results that are currently saved in the Site(s).

© Copyright 2020 Softing IT Networks

35

WireXpert User Interface

Select Custom Limits to import custom limit files from an USB drive. Ensure custom limit files are stored
in the “CustomLimits” folder in the USB drive before inserting to WireXpert. Select custom limit file(s) tobe loaded and select [Copy Custom Limits].
User Guides
Refer “User Guide – Copper Custom Limits” for more information on creating
and using custom limits.
Select Label List to import label list created with eXport PC software using an USB drive. Ensure custom
limit files are stored in the “Label List” folder in the USB drive before inserting to WireXpert. Select label
list file(s) to be loaded and select [Copy Label files].
User Manuals
Refer “User Manual – eXport” for more information on generating list-basedlabels.
Select Activate License(s) to load license(s) to upgrade WireXpert 500 to WireXpert 500-plus or
WireXpert 4500. Ensure custom limit files are stored in the “Features” folder in the USB drive before
inserting to WireXpert. Select license key(s) to be activated and select [Activate License(s)]. License(s)are unique to devices and cannot be transferred between units.
User Guides
Refer “User Guide – License Upgrade” for more information on upgrading
WireXpert through licensing.
4.2.2.5.9 Battery
Press the [SETUP] button  System Settings  Settings 2  Battery to view battery level. Battery
levels of LOCAL and REMOTE will be displayed if units are connected.

36© Copyright 2019 Softing IT Networks

WireXpert User Interface

4.2.2.5.10 Restore Factory Settings
Press the [SETUP] button  System Settings  Settings 2  Restore Factory Settings to restore
WireXpert to factory settings. Click [OK] to begin restoration.

CAUTION
Restoring WireXpert to default settings is an irreversible action. All user definedsettings and stored test results will be erased and cannot be recovered.
4.2.2.5.11 Touch Screen Calibration
Press the [SETUP] button  System Settings  Settings 2  Touch Screen Calibration to re-adjust
mechanical alignments of the touch screen. Follow on-screen instructions to begin calibration. Screen
can only be exited after the screen calibration is completed.

4.2.2.5.12 Device Info
Press the [SETUP] button  System Settings  Settings 2  Device Information to check the device
and probe serial numbers, software and hardware version numbers, device last calibrated date andprobe test count.
© Copyright 2020 Softing IT Networks

37

WireXpert User Interface

Click on the [Device Info] button for more information on the firmware.

Device firmware build version

Note
To ensure accurate test results, it is recommended to have WireXpert units calibrated
every year. Contact vendor for more information.

4.2.3 The [DATA] buttonmode.

The [DATA] button provides archive and data management ability to saved sites and test
results. Saved test results can be renamed or deleted in this option. Select Copper  [View] toview saved copper test results. Saved copper test results can only be viewed in copper test

View Results

38

Manage Results© Copyright 2019 Softing IT Networks

WireXpert User Interface

CAUTION
Deleting a site will also delete its containing test results.

4.2.4 The [TOOLS] button
The [TOOLS] button provides advanced options for in-depth troubleshooting and expert
WireXpert users. These options include; -Requires:
LOCAL and REMOTE1
Set Reference – Performs loss
compensation throughcalculation from setting ofreferencing point.
Requires:
LOCAL and REMOTE
Learn NVP – Performs a test to
determine the NVP of thecable using the length value3
programmed by the user.
Requires:
LOCAL and REMOTE
Factory Calibration – Performs
device calibration andconfigurations. For Service
Centre use only.

Requires:1. LOCAL only
2. Wi-Fi dongle#
Wifi – Enables WireXpert toconnect to the Wi-Fi to
perform eXport Cloudfeatures.

© Copyright 2020 Softing IT Networks

Requires:
LOCAL or REMOTE2
Length & Delay –Performs atest to determine the lengthand propagation delay of eachpair using the NVP value
programmed by the user.
Requires:
LOCAL or REMOTE

Requires:
LOCAL and REMOTE
Wiremap – Verifies end-to-end
wire mapping of cable.

Requires:
LOCAL or REMOTE

Reset Probe – Resets
Permanent Link adapter usage
count once a new link cord isin place.

About – Displays worldwide
contact and End User License
Agreement (EULA)
information.

Requires:1. LOCAL or REMOTE
2. Tone Generator#
Tone Generator – Detects farend of open-ended cables
when paired with Tone
Generator. See “ToneGenerator mode” for more
information.

Requires:1. LOCAL & REMOTE
2. License key(s)#Licensing – Performs device
upgrade from WireXpert 500
to WireXpert 500+ or/to
WireXpert 4500.

Requires:1. LOCAL only
2. eXport Cloud#eXport Cloud – Enables
WireXpert to download List
Based Testing (*.LBT) andupload project site (*.WSD)
files to the eXport Cloud
platform. Active network
connection is required.

39

WireXpert User Interface

Note
LOCAL and REMOTE - Action can only be performed when both LOCAL and
REMOTE units are connected.
1

LOCAL or REMOTE - Action can be performed using either the LOCAL or REMOTE
unit. End-to-end connection is not required.
2

3

30 meters (100 feet) cable length is at least required to get a valid NVP value.

Optional purchase required for feature. Contact local vendor for more
information.
#

User Manuals
Refer “User Manual – eXport Cloud” for more information on installing and using
eXport Cloud.
User Manuals
Refer “User Guide – License Upgrade” for more information on upgrading
WireXpert 500 to WireXpert 500+ or/to WireXpert 4500.

40© Copyright 2019 Softing IT Networks

Setting Reference

Setting Reference
It is necessary to perform a set reference measurement if the LOCAL and REMOTE units are being paired
for the first time. If there is a mismatch in firmware versions or a set reference has not been performed,
AUTOTEST will be denied, until these conditions are corrected. Please also verify before testing thatcalibrations are not expired.
Note
Diagrams and images used are for illustration purposes only and do not represent
suggested test values. Reference and test values vary to usage and condition.
To set reference,
1. Connect a Permanent Link adapter to the LOCAL unit.
2. Connect a Channel adapter to the REMOTE unit.
3. Connect the 2 units together using the patch cord provided in the kit.
4. Press the [TOOLS] button  Set Reference

5. Set Reference will fail in the event of•

Adapter probe mismatch, i.e., two channel or permanent link adapters•

Firmware version mismatch between LOCAL and REMOTE units•

No connection between LOCAL and REMOTE units© Copyright 2020 Softing IT Networks

41Configuring an AUTOTEST

Configuring an AUTOTEST
6.1 Setting up WireXpert
After configuring the system settings, follow these steps to set up an AUTOTEST.
1. Press the [SETUP] button  Project Settingsa. Site – Create or select a Site
b. Operator – Create or select an Operatorc. Label Source – Select cable labelling scheme. Load labels from USB flash drive if using List
Based Testing (LBT).
d. AutoSave – Enable option for WireXpert to auto save every PASS result.
2. Press the [SETUP] button  Quick Setup to select pre-defined test limit settings using matching
generic UTP/FTP cable and connector.
3. Manual Setup – Press the [SETUP] button  Test Settings
a. Test Limit – Select a test limit
b. Cable – Create a custom or select cable from list. Select [Generic] if unsure.
c. Connector - Create a custom or select connector from list. Select [Generic] if unsure.
d. Test Options  General

42

i.

Locator – Enable option to display the worst NEXT and Return Loss (RL) fault
information in an AUTOTEST result. Default has been set to OFF.

ii.

Cable Pairing Type – Select if cable under test is paired using the T568-A or T568-B
standard. Option will be disabled when a TERA Channel or TERA Link adapter isconnected to WireXpert.

iii.

Autotest on Connection – Enable option to perform AUTOTEST on, i.e., as soon as theconnection between LOCAL and REMOTE units is established. Default option has been
set to OFF.

iv.

Direct Attach – Enable option only if performing Direct Attach installation test. Default
option has been set to OFF.

v.

AC Wire map – Enable option to bypass testing DC parameters. Default option has been
set to OFF.

vi.

Crossover – Enable option and select wiremap(s) if cable under test is a crossover cable.
Option is only available when Application Standards is selected as the Limit Standard.
© Copyright 2019 Softing IT Networks

Configuring an AUTOTEST

Default option has been set to OFF.

Straight-through cable

Crossover at pairs 12, 36

Crossover at pairs 12, 36, 54, 78

Crossover at pairs 12, 36 and reversed atpairs 54, 78

Crossover at pairs 12, 36 for 2-pair cable

Please ensure you have the following components before conducting the test;•

WireXpert, LOCAL&REMOTE units (WX4500 or WX500-CU)

For Permanent Link Testing•2 x Permanent Link Adapter (WX_AD_6APL2)•

2 x Patch Cord (WX_AC_6ALCORD2)

For Channel Testing•

2 x Channel Adapter (WX_AD_6ACH2)
CAUTION
Do not connect WireXpert to a voltage source such as an active telephone jack.
Excessive voltage will damage the units and the adapter and void the warranty.

6.2 Testing Guide for Permanent Link Testing
Permanent Link (PL) test is comprised of the connection from a patch panel to a telecommunication
outlet (horizontal cabling) using Permanent Link test adapters at each end of the link under test.
1. Insert the Permanent Link adapters into the LOCAL and REMOTE units of the WireXpert.
2. Power ON WireXpert. Check that WireXpert is in Copper testing mode.
3. Set reference is required if the units are being paired for the first time.
4. Choose the Permanent Link limits and configure other settings if necessary.
5. Connect the LOCAL unit to the patch panel and REMOTE unit to the outlet using the patch cord
provided.
6. Press the [AUTOTEST] button to begin AUTOTEST.

© Copyright 2020 Softing IT Networks

43Configuring an AUTOTEST
PL adapter

Link under test

Patch Panel

Horizontal Cabling

PL adapter

OutletLOCALREMOTE

6.3 Testing Guide for Channel Testing
Channel (Ch) test is comprised of the connection from an active device (eg. Router) in a data rack, atelecommunication outlet (horizontal cabling) and the connecting patch cords at both ends using
Channel test adapters at each end of the link under test. The recommended length for the patch cordfrom the patch panel and the outlet is 5m.
1. Insert the Channel adapters into the LOCAL and REMOTE units of the WireXpert.
2. Power ON WireXpert. Check that WireXpert is in Copper testing mode.
3. Set reference is required if the units are being paired for the first time. Refer to Chapter 2 on how to
set reference.
4. Choose the Channel limits and configure other settings if necessary.
5. Connect the LOCAL unit to the patch panel and REMOTE unit to the outlet using the patch cordconnected to be used.
6. Press the AUTOTEST button to begin AUTOTEST.
Ch adapter

Link under test

Patch Panel

LOCAL

44

Horizontal Cabling

Ch adapter

Outlet

REMOTE© Copyright 2019 Softing IT Networks

Performing an AUTOTEST

Performing an AUTOTEST
7.1 Understanding AUTOTEST
Press the [AUTOTEST] button once settings and limits have been selected. WireXpert will use the lastconfiguration or factory settings to perform the AUTOTEST if new settings are not configured.
WireXpert will display summarized result with PASS or FAIL once AUTOTEST is completed. Press the[View] details button to view the comprehensive result or the [Save] button to save the results.

AUTOTEST in progress

AUTOTEST complete[View] results

Click on the parameter to display a more comprehensive individual result.
In plot view, press the [Worst Margin] button to view the worst margin. Press again to view the [Worst
Value] or press the [Manage] button for more options.

Wire map© Copyright 2020 Softing IT Networks

Length and Delay

Resistance45

Performing an AUTOTEST

46

Insertion Loss

Return Loss

Near-end Crosstalk

Attenuation to Crosstalk Ratio,
Near-end (ISO only)

Attenuation to Crosstalk Ratio,
Far-endPower-sum NEXT© Copyright 2019 Softing IT Networks

Performing an AUTOTESTPower-sum ACRN (ISO only)

Power-sum ACRF

Length & Delay

7.2 NEXT and RL Fault Locating
If the Locator option is enabled in the Test Settings  Test Options, WireXpert will display extremely
critical troubleshooting information in the form of a Time Domain Reflection plot for NEXT and Return
Loss.
Near-End Crosstalk (NEXT) is an undesired effect of interference between two pairs in a cable measured
at the same end of the cable as the interfering transmitter. In crosstalk, the signals traveling throughadjacent pairs of wire in twisted-pair cabling interfere with each other. Cross talk is undesirable.
The NEXT Locator displays the locations where crosstalk is excessive in the cable and indicates the exact
location in meters or feet from the LOCAL unit of where high crosstalk issue is seen. View individual plotby selecting/deselecting cable pairs. Each two pairs of cables that have been tested for crosstalk arerepresented by different color in the plot.
Return Loss (RL) is the measurement (in dB) of the amount of signal that is reflected back toward thetransmitter. Impedance mismatches and discontinuities caused by terminations to plugs or jacks andwithin the plug/jack connection itself contributes to RL. Excessive bends and damaged cable will also
contribute to RL. The greater the variation in impedance, the greater the return loss reading.
The RL Locator displays the locations where return loss occurred during an AUTOTEST and indicates theexact location of where the highest return loss occurs. View individual plot by selecting/deselecting
cable pair. Each pair of cables that has been tested is represented by different color in the plot.
Each peak result in NEXT and RL Locator could represent a new joint connection, rarely a coupler andnot necessary a bad cable that requires replacement.

© Copyright 2020 Softing IT Networks

47Performing an AUTOTEST

Locator – To enable, press the[SETUP] button  Test
Settings  Test Options 
LocatorNEXT Locator

Return Loss Locator

7.3 Managing test result(s)Test results can be manually saved by pressing the [Save] button after an AUTOTEST is completed.
When prompted, enter label name and click [OK] to save.

The [Save] icon will disappear
once saving is completed.

Test results will be named in alpha-numeric sequence with reference to the selected Label Source([SETUP] Project Settings  Label Source) when AUTOTEST is completed.
PASS results will automatically be saved if the “Auto Save” option is enabled. FAILED results will besaved manually.

48© Copyright 2019 Softing IT Networks

Performing an AUTOTEST

To view saved results,
1. Press the [DATA] button.
2. Select [Copper] and press the [View] button.
3. Select the test results click [View] to view results.
4. Select the next page for more results.
To delete a saved result,
1. Press the [DATA] button.
2. Select [Copper] and press the [View] button.
3. Press the [Manage] button.
4. Select result(s) and press the [Delete] to delete result(s).
To rename a saved result,
1. Press the [DATA] button.
2. Select [Copper] and press the [View] button.
3. Press the [Manage] button.
4. Select result and press the [Rename] to rename result.

© Copyright 2020 Softing IT Networks

49Performing an AUTOTEST

View results

Manage results

7.4 Exporting test results into eXport PC SoftwareeXport is a data management software designed to work seamlessly with WireXpert. Saved results canbe exported using the following methods;

7.4.1 USB Flash Drive
1. Insert USB flash drive to USB port on WireXpert.
2. Device will automatically detect flash drive.
3. Otherwise, press the [SETUP] button  System Settings  Settings 2  Storage  USB
4. Select [Test Results].
5. Click [OK] to proceed.

6. Launch eXport software.
50© Copyright 2019 Softing IT Networks

Performing an AUTOTEST

7. Go to File  Create New Project to create new project space.

8.

Go to Import  USB Flash Drive and select the flash drive. Click [Import].

9. Select database and click [Import Selected] to import or click [Import All] for all databases.

© Copyright 2020 Softing IT Networks

51Performing an AUTOTEST

7.4.2 Direct connection via USB cable
1. Connect WireXpert to your PC using a mini USB cable.
2. Launch eXport software.
3. Go to File  Create New Project to create new project space.
4. Go to Import  Device.

5. Select database and click [Import Selected] to import or click [Import All] for all databases.

User Manuals
Refer “User Manual – eXport” for more information on how to use the software.

52© Copyright 2019 Softing IT Networks

Declarations

Declarations© Copyright 2020 Softing IT Networks

53Declarations

This device complies with the requirements of the EC directive 2004/108/EG "Electromagnetic
Compatibility" (EMC directive). It meets the following requirements:

Note
A Declaration of Conformity in compliance with the above standards has
been made and can be requested from Softing Singapore Pte Ltd.

China ROHS
The WireXpert device and its test components are China ROHS compliant.

WEEE
Electrical and electronic equipment must be disposed of separately fromnormal waste at the end of its operational lifetime.
Please dispose of this product according to the respective national
regulations or contractual agreements. If there are any further questions
concerning the disposal of this product, contact Softing IT Networks.

CAUTION
This is a Class A product. In a domestic environment this product may causeradio interference. In that case the user may be required to take adequate
measures!

ROHS
The WireXpert device and its test components are ROHS compliant.

ETL Intertek Verified
WireXpert device is ETL verified to ANSI/TIA IIIe, IEC 61935-1 levels IIIe & IV
and currently proposed Level V draft, with the applicable measurement
accuracy.

Class 1 Laser Product
The light source transmitted from the following fiber test modules – Single
Mode (SM), Multi-Mode (MM) and Encircled Flux compliant Multi-Mode(MMEF) are classified as Class 1 lasers and are very low risk and "safe underreasonably foreseeable use", including the use of optical instruments for
intrabeam viewing.

54© Copyright 2019 Softing IT Networks

Declarations

Class 1m Laser Product
The light source transmitted from the following fiber test modules – MPO
REMOTE are classified as Class 1m lasers and have wavelengths between
302.5 nm and 4000 nm and are safe except when used with optical aids.

© Copyright 2020 Softing IT Networks

55

Appendix

156[Add]– Adds a site, operator
or customised cable,
connector to the database.

[Retest] – Performs an
AUTOTEST on selected resulton List-Based Testing.

[Back] – Returns to previous
screen. Unsaved options will
be discarded.

[Cancel] – Discards option.

[Delete] – Deletes a site,
operator or customised
cable, connector from thedatabase.

[View]– View selected result.

[Manage] – Enables[Rename] and [Select all]options.

Format, Reset – Performs
non-reversible setting/data
restoration to factory
defaults.

Forward – Proceed to thenext screen.

[Device info] – Displays
device firmware build
information.

[OK] – Confirms and saves
current option.

[Rename] – Renames saved
test result in the DATA
menu.

[Restart] – Restarts current
procedure.

[Save] – Saves current test
result. Icon will disappear
after a successful save.

[USB] – Reads USB flash
drive to perform upgrade
firmware, test result export
or custom limit and label list
import© Copyright 2019 Softing IT Networks

Appendix[Select all] – Selects all dataon screen.

[Next pair] – View next pairof plots of the current result.

Fibermap – Displays
mapping of fiber being
tested.

[MPO chart/grid] – Toggles
between displaying of power
loss in bar chart or gridformat for fiber testing.

[Scope live/test] – Toggles
between ‘Live’ and ‘Test’mode when inspecting a
SM/MM fiber with theinspection scope.

[Set Reference] – Performs
result referencing between
the LOCAL and REMOTE
units.

[Transmit ON/OFF] –Toggles between enablingand disabling of light sourceon the REMOTE unit.

2© Copyright 2020 Softing IT Networks

57

Glossary
AC Wire Map: Feature in WireXpert
that enables test runs measurementswith Power over Ethernet (PoE)
midspan devices in between. WireXpert
supports cable runs with IEEE 802.3 af
and 802.3 at injectors. Recommended
to disable when not performing a test
through a PoE device.
ACRF: Acronym for Attenuation to
Crosstalk Ratio Far-end, formally known
as Equal Level Far End Crosstalk(ELFEXT), is a calculated parameter,
derived from subtracting theattenuation of the disturber at the far
end pair from the FEXT on the
disturbed pair. This is expressed in dB.
ACRN: Acronym for Attenuation to
Crosstalk Ratio Near-end, formallyknown as ACR, is a calculatedparameter, derived from subtracting
the attenuation of the disturber pairfrom the NEXT on the disturbed pair.
This is expressed in dB. Test parameter
is not required for TIA cabling
standards.
Adapter: The modular test probe usedwith WireXpert when used in thisManual’s context, unless otherwise
stated.
CAT: Abbreviation for “Category”,
describes the transmission
performance and construction ofcopper cabling links associated with
ANSI/TIA-568 Standards.
Channel: Definition of cabling elements
in horizontal cabling, permanently
installed data cable including patch
cords/jumpers at each end.
Class: Describes the transmission
performance and construction ofcopper cabling links associated with
ISO/IEC 11801 Standards.
Crosstalk: Disturbance caused by theelectric or magnetic fields along a cablepair. It can occur when removing too
much of the pair shield or untwistingpairs too much during termination,

58

using the wrong components, selecting
wrong settings or using inferior quality
test setups.
Crossover Cable: Cable used to connect
two devices of the same type, e.g., two
computers or two switches. Example, acrossover cable can have the following
A-to-B ends pin-outs: 1-3, 2-6, 3-1, 4-7,
5-8, 6-2, 7-4, 8-5.

Crossed Pairs: Generally referring towiring fault, also known as Transposed
Pairs, is a twisted pair connected tocompletely different pin pairs at bothends.

E2E: Acronym for End-to-End. Generally
referring to End-to-End links.
Application that requires directconnection to machines or machines to
network devices in the industrial
environment, potentially containing 1
or multiple segments with 2 or multiple
connections in the link.
EULA: Acronym for End User License
Agreement. Starting from v7.3, user
will require to read and agree to theterms and conditions stipulated by
Softing before proceeding to use
WireXpert and eXport PC firmware.
eXport: Project management PC
software to seamlessly work with
WireXpert. Required to upgrade device
firmware. Not to be confused withactions “file export”.
FTP: Acronym for Foil-screened Twisted
Pairs. See F/UTP. Not to be confusedwith “File Transfer Protocol”.

Crossover cable is an example ofcrossed pair cable, but not crossed paircables are crossover cables.
dB: Acronym for decibel, is the ratio
unit used to measure the power levelof an electrical signal by comparing itwith a given level on a logarithmic
scale.
DCRU: Acronym for Direct Current
Resistance Unbalance, is themeasurement of DC resistance
between and in pairs of twisted pair
LAN cabling.

F/FTP: Cable with overall foil shieldwith FTP. Commonly used in 10GBaseT
applications.

F/UTP: Cable with an overall foil shield(F) with UTP. Commonly used in
10GBaseT applications. Also known as
FTP.

DCS: Acronym for Dual Control System,
is a unique feature of WireXpert thatprovides real-time graphical test dataon both the LOCAL and REMOTE units.

GUI: Acronym for Graphical User
Interface, generally referring to thetouchscreen interface on WireXpert inthis Manual’s context, unless otherwise
stated.

Delay Skew: Refers to test used todetermine the difference inpropagation delay between the fastestand slowest set of wire pairs. In
general, the lower the skew value, thebetter.

IL: Acronym for Insertion Loss, is theloss of signal power resulting from theinsertion of jack to a plug. This isexpressed in dB.

Device: Refers to WireXpert unit in thisManual’s context, unless otherwise
stated.

In-Pair: Generally referring to effects
within a pair of wires (eg., pins 1 & 2) intwisted pairs cable in this Manual’scontext.

© Copyright 2019 Softing IT Networks

Glossary
ISO: Acronym for International
Organization of Standardization.
Generally referring to ISO/IEC 11801
Standards for generic cabling for
general purpose telecommunication
cabling systems or structured cabling.
Jack: Generally referring to a connector
with female electrical contact orsocket.
LBT: Acronym for List-Based Testing, isa unique feature of WireXpert thatallows creation of label list and perform
AUTOTEST in a hierarchical format.
Limit: Or test limit, refers to theperformance threshold that a cableinstallation is obliged to meet.
Link: See Permanent Link.
LOCAL: Refers to the WireXpert unit
labelled “LOCAL”.
Margin: Refers to the difference
between the selected test limit value
and the measured value. This isexpressed in dB. In general, a higher
positive margin value represents agood test result, a negative value
represents a FAIL test, and zero whenthe measured value is the same as thetest limit.
Marginal: Generally referring tomarginal PASS/FAIL, refers to thetested value’s margin is smaller than
the accuracy for the test parameter.
NEXT: Acronym for Near-End Crosstalk,is the measurement of the ability of acable to reject crosstalk. In general, thehigher the NEXT value, the greater therejection of crosstalk at the local
connection.
NVP: Acronym for Nominal Velocity of
Propagation, is the speed at which thedata signals travel down the cable
expressed as a percentage of the speedof light in a vacuum. NVP is required todetermine the length of a cable during
a test.
P2P: Acronym for Pair-to-Pair, generally
referring to effects between two pairs© Copyright 2020 Softing IT Networksof wires (eg., 1,2 & 3,6) in twisted pairs
cable in this Manual’s context.
Patch Cord: Also known Patch Cable, isan electrical cable used to connect one
electronic to another for signal routing.
Commonly produced in different colors,
and are usually not longer than 2
metres.
Patch Panel: Also known as Patch Bay,is a device or unit featuring a numberof jacks, for connecting and managing
incoming and outgoing LAN cables.
Plug: Generally referring to a connector
with male electrical contact or pin.
Permanent Link: Definition of cabling
elements in horizontal cabling,permanently installed data cable.

PSNEXT: Acronym for Power Sum NEXT,is the sum of NEXT values from threewire pairs as the affect the other wire
pair. There will be four PSNEXT results
at each end of the link tested. In
general, the higher the PSNEXT value,
the lower the crosstalk.
REMOTE: Refers to the WireXpert unit
labelled “REMOTE”.
Reference Cord: Generally referring to
the patch cord used for settingreference prior to performing an
AUTOTEST.
Reversed Pairs: Generally referring towiring fault, a twisted pair is notconnected with straight-through pin-topin continuity. (eg., Pins 1 & 2 arereversed.)

PoE: Acronym for Power over Ethernet,
is the technology for wired Ethernet
LANs that allows electrical current
required for operation of devices to be
carried by data cables.
Probe: Generally referring to a device
used to connect a test equipment to a
device under test, e.g., the Channel
adapter and Video Inspection Scope areprobes to the WireXpert.

RL: Acronym for Return Loss, is ameasurement of reflections caused by
the impedance changes at all locations
along the link. This is expressed in dB.

Propagation Delay: The measurementof how long it takes for the signal to
travel down the cable being tested.

Set Reference: The compensation for
the effects of ambient conditions, e.g.,ambient temperature, in the LOCAL and
REMOTE units, to ensure maximum
measurement accuracy of WireXpert.

PSACRF: Acronym for Power Sum ACRF,
formally known as Power Sum Equal
Level Far-End Crosstalk (PSELFEXT), is
the difference between the test signal
and the crosstalk from the other pairs
received at the far end of the link. Thisis expressed in dB.
PSACRN: Acronym for Power Sum
ACRN, formally known as Power Sum
Attenuation to Crosstalk Ratio (PSACR),
is the difference between each wire
pair’s attenuation (IL) and the
combined crosstalk received from theother pairs. In general, higher PSACRN
values corresponds to better cabling
performance. This is expressed in dB.
Test parameter is not required for TIA
cabling standards.

Shielded: Generally referring to anelectrical cable or connector being
enclosed in or by conductive layers toprovide noise screening.
Shield Test: Also known as Shield
Integrity Test, refers to theidentification of a shield being present
and connected at both ends on theactual cable under test.
Single Ended Test: Feature in
WireXpert, referring to the ability todetermine the loss of copper cable at
one end without the use of the
REMOTE unit.
Split Pairs: Generally referring to wiring
fault, individual wires from two59

different twisted pairs are connected toa set of pins that are supposed to beconnected to a real pair of wires. The
fault usually happens when the samewires are swapped on both sides of thelink. (eg., Pins 3 & 5 are swapped on
both sides of the link.)

S/UTP: Cable with an overall braid
screen (S) with unscreened UTP. Also
known as STP.

STP: See S/UTP.
S/FTP: Cable with an overall braid
screen (S) with FTP. The “shield”
underneath the jacket is a braid, andeach individual pair is surrounded by its
own foil barrier. The additional foil on
the individual pairs limit the amount ofcrosstalk between them. Also known as
S-STP.

SF/UTP: Cable with both an overall
braid screen (S) and foil shield (S) with
UTP. The overall braided screen is very
effective at protecting EMI fromentering or exiting the cable. Also
known as S-FTP.

3-3, 4-4, 5-5, 6-6, 7-7, 8-8.

T568A/B: T568A and T568B are twocolor codes approved by ANSI/TIA/EIA
that are used for wiring eight-position
RJ45 plugs.

TIA: Acronym for Telecommunications
Industry Association. Generally
referring to TIA-568 Standards fortelecommunication products andservices.
TIA-606-A: The guidelines and choicesof classes of administration formaintaining telecommunications
infrastructure as defined by the TIA
Standards.
Unshielded: Generally referring to anelectrical cable or connector that are
not enclosed in or by conductive layers.
U/FTP: Cable without overall shielding
or braid (U) with FTP. Commonly usedin 10GBaseT applications.

UTP: Acronym for Unscreened Twisted
Pairs. See U/UTP.
U/UTP: Cable without shielded with
UTP. Also known as UTP.

Straight Through Cable: Cable used toconnect computers to hubs or switches.
A straight through cable has thefollowing A-to-B ends pin-outs: 1-1, 2-2,

60

Wiremap: The parameter that checks
the correct termination of cable wires
in telecommunication outlets and patch© Copyright 2019 Softing IT Networks

Related documents

Related documents
Application Note – E2E Link Test
Application Note – MPTL
Quick Start Guide – Copper Certification Testing
Quick Start Guide – Fiber Certification Testing
Quick Start Guide – Encircled Flux Compliant Multimode Fiber Certification Testing
Quick Start Guide – MPO Certification Testing
Quick Start Guide – Digital Fiber Inspection Kit
User Manual – Fiber Certification Testing
User Manual – MPO Certification
User Manual – eXport
User Manual – CAT6A+ Adapters
User Guide – List Based Testing
User Guide – Installing eXport PC Software
User Guide – License Upgrade
User Guide – eXport Cloud
User Guide – Custom Limits
User Guide – Tone Generator© Copyright 2020 Softing IT Networks

61Technical Support
Softing’s global presence ensures our customers receives sales and technical support anywhere around theworld. For more information: https://itnetworks.softing.com

The Americas
Softing Inc.
7209 Chapman Highway
Knoxville, TN 37920
Phone: +1 865 251 5252
E-mail: sales@softing.us
Asia / Pacific
Softing Singapore Pte. Ltd.
73 Science Park Drive#02-12/13, Cintech I
Singapore Science Park 1
Singapore 118254
Phone: +65-6569-6019
E-mail: asia-sales.itnetwork@softing.com
Europe/Middle East/Africa
Softing IT Networks GmbH
Richard-Reitzner-Alle 6
D-85540 Haar, Munich
Phone: +49 89 45 656 660
E-mail:info.itnetworks@softing.com
Softing Italia Srl.
Via M. Kolbe, 6
20090 Cesano Boscone (MI)Phone: +39 02 4505171
E-mail: info@softingitalia.it

62

Softing Shanghai
Room 416, 4/F, Mytech Intelligence Park,
No.1999 East Jinxiu Road, Pudong District,
201206, Shanghai, China
Phone: +86 (21) 61063031
E-mail: china-sales.itnetwork@softing.com
Softing SRL
87 Rue du Général Leclerc
Creteil, Île-de-France
94000 (Paris)
Phone: +33 1451 72805
E-mail: info.france@softing.com
Austria
Buxbaum Automation GmbH
Eisenstadt
Phone: +43 2682 7045 60
E-Mail: office@myautomation.at© Copyright 2019 Softing IT Networks