INSTRUCTION MANUAL Model: GDCS-125H
Inside Measurements OnlyPrecision Measuring Tools

DIGITAL HOOK CALIPERS■PARTS IDENTIFICATION & FUNCTION●Use care to avoid damaging
terminals when insertingbattery.

⑦Scale②Jaw(Hook)
Terminals①Measuring
Surfaces

Hook

Slide In

Terminals①Loosen the Lock Screw and turn power ON.
②With jaws closed, press the [ZERO] Button.
③When “0.00mm” is displayed on LCD, Caliper is ready tostart measurement.

■Comparison Measure Mode

①Loosen the Lock Screw and turn power ON.
②With jaws open to desired reference position (any point isOK,) press [ZERO] Button.
③When “0.00mm” is displayed on LCD, Caliper is ready tostart measurement.

BatteryPressed Down

⑥ Output Connector③LCD Display■Absolute Measurement ModeBattery
Cover●Terminals may be broken if battery is pressed straight down.
●Insert battery at angle, underneath the small hook, as shown.

⑤ [HOLD] Button④ Lock Screw①To turn on, press the [ON/OFF ] Button.
②To turn off, press the [ON/OFF ] Button.
③With power on, press the [HOLD] Button to fix the value onthe display.

vemo● For safe and proper use, please read this instruction manual before use and follow the procedures described.
Please keep manual where it is accessible to user for future reference.
● Keep this manual with the instrument if transferred or leased to a third party.
● For inquiries about this product, please contact dealer or Niigata Seiki at the address listed on the following page.

Notice Use only CR2032 typebattery.

HOW TO USE■Power ON/OFF①Slide cover in direction of arrow.
②Insert battery with (+) side outward.
③Replace battery cover.

ReThank you for purchasing Niigata Seiki Digital Hook Calipers.
This is a precision measuring tool with a narrow jaw for measuring inside dimensions in tight spaces.

INSERTING・REPLACING BATTERYBattery⑧Battery Cover⑨ [ ZERO ] Button

AFTER USE CARE・STORAGE⑩ [ ON/OFF ] Button①Measuring Surfaces … Makes contact with object being measured.
②Jaw(Hook) …………… Position on the part being measured.
③LCD Display ………… Displays digital measurement.
④Lock Screw ………… Tighten to prevent the slide and display
from moving.
⑤[ HOLD ] Button …… When pressed, "HOLD" Icon appears onDisplay and measurement will be fixed.

SAFETY PRECAUTIONS
To prevent harm to yourself or others, and to prevent damage toproperty, always follow the procedures marked by the followingsymbols.

⑥Output Connector … Connect to optional Bluetooth Box
to send measurements to PC.
⑦Scale ………………… Supports LCD Display Unit.
⑧Battery Cover ……… Open to access battery.
⑨ [ ZERO ] Button … When pressed, Display will read "0.00"⑩[ON/OFF] Button … For turning power ON / OFF

Denotes a requirement-You MUST do

WARNING: Indicates risk of personal injury or property damage if not followed.
Read the manual and follow the directions.
・Use of product other than as described in this manual
may cause accident.
Use only for measuring.
・Use for any purpose other than measuring may
damage or wear the instrument. Improper use may
also cause accident.
Handle with Care.
・Do not drop or subject instrument to excessive shock.
Do not place under heavy objects. Improper handling
may cause damage or poor accuracy.
・Do not scratch instrument, for example by writing ID
number.
Avoid using in high electric fields, such as nearfluorescent lights or switching power supplies.
・Incorrect value may be displayed on LCD

PREPARATION FOR USE●Wipe off rust proofing and dirt from measuring surfaces
and slide.
Contamination of surfaces may cause measurement error.

Use and store only in environment meeting thefollowing conditions:●Dry location protected from rain, water, and oils.
●Location that is not excessively hot.
●Location protected from use by children andunauthorized people.
・Use in places contrary to the above may cause
damage to the product resulting in accident or injury.
Jaw Tips Are Sharp̶Handle With Care.
・Careless handling may cause injury.
Do not disassemble or modify.
・Disassembly may damage product or cause accident.
For repair, please contact distributor or place of purchase.
Do not write on instrument with electric engraver.
・Do not engrave instrument body or jaw, such as an ID
number with engraver or electric pen.

Before using the caliper, please follow the steps below:●Allow the caliper and object to be measured time to reachthe same temperature.
A temperature difference between the object to be measured
and the caliper may cause measurement error. Allow enoughtime for the temperatures to equilibrate.

Workpiece

For rust prevention, lightly apply anti-rust oil, or place in rustpreventive bag.

Measurement in tight location●Store with Lock Screw loosened.

If stored with Lock Screw tight, expansion due to temperaturechanges may cause excessive force which will cause inaccuracy.

●Store in a cool, dry, and dark location in theprovided case.

Keep out of direct sunlight and moisture, and please keep
secure from unauthorized personnel.

Please ObserveDenotes a prohibition-You MUST NOT do●After use, wipe off any dirt or oil from measuring
surfaces, sliding surfaces, and exterior using adry cloth and perform rust prevention treatment.

●Remove battery if Caliper will not be used for along period of time.
Batteries may leak if left in unused instrument for an extendedperiod.

TROUBLESHOOTING
Ｑ．Nothing is displayed on LCD.
Ａ．The battery may be running low, please replace battery
with a new one.
Ｑ．With power on, the number on the LCD does notchange as Display Unit is moved.
Ａ．Please reset by removing battery for about 30sec., andthen replacing.
Ｑ．Nothing happens when buttons are pressed.
Ａ．Make sure battery is properly inserted.
Ｑ．The characters “MIN”, “MAX”, or “△” are displayed on LCD.
Ａ．Cycle power, or apply several long presses to [HOLD]
Button. Use as normal once character is off.。
Ｑ．The correct value is not displayed.
Ａ．High humidity may cause dampness. Please dry off theinstrument and use only in dry location.

SPECIFICATIONS●Measurement Range ……… 4 -125mm●Resolution…………………… 0.01mm●Maximum Permissible Error … ±0.03mm※Not including quantization error (±1 count)●Display ……………………… 5 Digit LCD●Auto Off ……………………… approx. 20 min●Maximum response speed… Slide in pos. dir.…1.5m/s
Slide in neg. dir.…0.75m/s※Left/Right as shown in diagram at top of manual.

●Operating Temp. Range ……
0°
C〜40°
C
●Storage Temp. Range ……… −10°
C〜60°
C
●Power ……………………… (1x) CR2032 Lithium Battery●Battery Life ………………… approx. 2.5 years in normal use●Weight ……………………… 125g●Accessory …………………… CR2032 Lithium Battery (1x for test)

SERVICE●If Instrument is not working properly, or if you have anyquestions, please contact distributor or place of purchase.
●Please note, manufacturer is unable to respond to inquires orprovide service directly. Please contact distributor or place ofpurchase.

Niigata Seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

取扱説明書

内側測定専用
精密測定工具

デジタルフックノギス品番

GDCS-125H⑤HOLDスイッチ⑥ 出力コネクタ③ LCD表示部●電池をセットする際は、端子の折れにご注意ください。

電池⑦本尺押し込む

①測定面

端子

⑧電池蓋
⑨ZEROスイッチ②ジョウ …………… 被測定物の内側を測ります。

⑥出力コネクタ ……… 別売の Bluetooth BOX を接続することで、測定値をパソコンへ出力することができます。

④止めねじ ………… デジタル液晶部のスライドを固定します。

⑧電池蓋 ……………… 開けて電池をセットします。

③LCD表示部 ……… 測定数値をデジタル表示します。

⑦本尺 ………………… デジタル液晶部のスライドを保持します。

⑤HOLDスイッチ … 押すと画面上に" HOLD"と表 示され 、表示している数値を固定できます。

⑨ZEROスイッチ …… 押すと画面数値が"0.00"と表示されます。

安全上のご注意⑩ON/OFFスイッチ … 電源のON/OFFを行います。

必ずお守りください

お使いになる人や他の人への危害、財産への損害を未然に防止するため、必ずお守りいただくことを右記のように説明しています。

してはいけない内容『禁止事項』です。

必ず実行していただく内容『強制事項』です。

お守りいただかないと『傷害を負う、または物的損害が発生するおそれがある内容』です。

取扱説明書をよく読み、指示に従う。
・取扱説明書に記載された内容以外での使用は、事故の原因となります。
測定作業にのみ使用する。
・指定用途以外へのご使用は、製品の破損や磨耗、予測できない事故の原因となります。
大切に取り扱う。
・ぶつける、落とすなどの衝撃を与えたり、重量物を上に乗せたりしないでください。精度不良や破損の原因となります。
・商品に傷をつけないでください。
インバータ式蛍光灯などの電界が発生する場所での使用は避ける。
・ＬＣＤ表示部の数値が誤作動を起こす原因となります。

使用前の準備

下記の条件を満たす環境で使用・保管する。
●雨や水、油などがかからない、乾燥した場所●高温にならない場所●子どもや、使用者以外が近付かない場所
・上記に反する場所での使用は製品の破損、事故やけがの原因となります。
ジョウ先端は尖っているので、扱いに注意する。
・けがの原因となります。
分解・改造を行わない。
・製品の破損や事故の原因となります。修理はお買い上げの販売店、または発売元までご連絡ください。
電気ペンなどで書き込まない。
・電気ペンなどで本尺やジョウなどに、番号などの書き込みをしないでください。

ご使用前に、下記の準備を行ってください●測定面、摺動面、外観などの汚れや防錆油などを拭き取る
そのまま測定を行うと誤差の原因となります。

ツメ①止めねじをゆるめ、電源を入れてください。
②ノギスのジョウを閉じた状態で "ZERO"スイッチを押してください。
③LCD表示部に"0.00mm"が表示されると計測開始ＯＫです。

■比較測定モード①止めねじをゆるめ、電源を入れてください。
②ノギスのジョウが開いた状態（どのポイントでもＯＫ）の時に"ZERO"スイッチを押してください。
③LCD表示部に"0.00mm"が表示されると計測開始ＯＫです。

入れ込む端子電池

使用後のお手入れ・保管方法⑩ ON/OFFスイッチ①測定面 …………… 被測定物の内側にこの面を当てて測定します。

■絶対測定モード

電池蓋●電池をセットする際に上から強く押し込むと、端子が折れて破損する場合があります。
●電池をセットする際は、右斜め方向から端子を外側に押すように、本体のツメの下へ入れ込んでください。

② ジョウ注意

電池は必ずCR2032をお使いください。

①電源をONにするには、”ON/OFF”スイッチを押してください。
②電源をOFFにするには、”ON/OFF”スイッチを押してください。
③電源ONの状態から“HOLD”スイッチを押すと数値が固定されます。

す外ご注意●正しく安全にお使いいただくため、ご使用の前に本取扱説明書を必ず読み、記載の手順に従ってご使用ください。
お読みになった後は、お使いになる方がいつでも見ることができる場所に大切に保管してください。
●第三者に譲渡・貸与される場合も、この説明書を必ず添付してください。
●当商品に関するお問い合わせは、お買い求めの販売店もしくは当社にご連絡ください。
④ 止めねじ■電源のON/OFF①電池蓋を矢印の向きに開けます。
②電池を、必ず（＋）側を表にしてセットしてください。
③電池蓋を元通りに装着してください。

この度は「デジタルフックノギス」をお買上げいただきありがとうございます。
この商品は、極小のジョウにより狭い箇所の内側測定が可能な精密測定工具です。

■各部の名称と機能

使用方法

電池のセット・交換●本体と被測定物の温度慣らしを十分に行う測定器本体と被測定物の間に温度差があると、誤差の原因となります。同じ温度になるよう、温度慣らしは十分に行ってください。

●測定面・摺動面・外観などの汚れや油などを乾いたウエスで拭き取り、防錆処理を行う
防錆処理は防錆油を薄く塗るか、防錆袋などに入れてください。

ワーク

狭い箇所の内側測定●止めねじをゆるめた状態で保管する止めねじを締めて固定すると、温度変化による膨張などで無理な力がかかり、精度不良の原因となります。

●付属のケースに入れ、乾燥した冷暗所に保管する

直射日光や湿気は避け、管理者以外が触れない状態で保管してください。

製品仕様●長期間使用しない場合は電池を製品から取り外した状態で保管する●測 定 範 囲：4 - 125mm●最 小 読 取 値：0.01mm●最大許容誤差：±0.03mm

トラブルと対応●表示：LCD 5桁表示
●オ ー ト オ フ：約20分●最大応答速度：右方向へのスライド…1.5m/s
左方向へのスライド…0.75m/s

長期間使用しないで放置すると、電池の液漏れの原因となります。

Ｑ．LCD表示部に何も表示されない。
Ａ．電池の残量が少なくなっている可能性があります。新しい電池と交換してください。

Ｑ．電源ONの状態でスライドさせてもLCD表示の数字が変わらない。
Ａ．一度電池を取り外し、30秒後に電池をセットし直してください。
Ｑ．スイッチを押しても何も反応しない。
Ａ．電池が正しくセットされているか確認してください。

※量子化誤差（±1カウント含まず）※右方向、左方向は各部の名称の図を正面にした場合●使用温度範囲：0℃〜40℃●保管温度範囲：−10℃〜60℃
源：リチウム電池 CR2032×1●電●電 池 寿 命：通常使用状態で約2.5年●質量：125g
品：リチウム電池 CR2032（テスト用）●付 属

Ｑ．LCD表示部に『MAX』『MIN』『△』などの文字が表示されている。
Ａ．一度電源を切るか、HOLDスイッチを何回か長押しして文字を消してからご使用ください。

Ｑ．正しい数値が表示されない。
Ａ．使用環境の湿度が高い可能性があります。本器を乾燥させ、湿度が低い環境下でご使用ください。

修理の際は●正常に作動しない場合や、不明な点がございましたらお買い上げの販売店または、発売元までご連絡ください。
●お問い合わせや、ご連絡が無いまま直接発売元に修理品などを送付されても処理、対応ができない場合がありますのでご了承ください。

発売元〒955-0055 新潟県三条市塚野目5丁目3番14号(0256)33-5502(代) FAX(0256)33-5528

URL http://www.niigataseiki.co.jp

I373-K1 2204