User Manual
Model No. WGC-150・WGC-200

Precision Measuring Instrument Holder

GRANITE COMPARATOR STAND■PART IDENTIFICATION③Fine Adjustment
Knob④StemMounting Hole

⑥Main Post

①Bracket①Bracket⑦Measuring
Platform

Please confirm the above items are included. If any items aremissing, please contact dealer or place of purchase.

⑥Main Post

■ASSEMBLY①BracketInsert①Bracket ････････････ Moves vertically with attachedmeasuring instrument.
③Fine Adjustment ･･･ Vertical fine adjustment of instrument
Knobclamp.
④Stem ･･････････････ Measuring instrument mounts here.
Mounting Hole⑤Instrument ･････････ Clamp secures measuring instrument.
Mounting Clamp⑦Measuring Platform･･･ Work surface.
Platform will wear with continued
use, so please use block gauge tocheck for wear before use.

Insert⑥Main Post ･････････ Holds bracket for vertical positioning.

Mount②Bracket Clamp ･････ Fixes bracket to main post.

②BracketClamp

５ Tighten⑥Main Post

Main Post Hole
⑦Measuring
Platform

WasherSpring Washer
M10 Bolt

Handle With Care.

・Do not drop or subject to shock -- instrument accuracy
may be affected.
・Do not allow water or oil to sit on measurement base,use a soft cloth to clean.

Use only for measuring.

・Use for any purpose other than measuring may damage orwear the instrument. Improper use may also cause accident.

Protect measuring platform from scratches.

・To prevent damage, use care when placing workpieces
onto platform surface, and avoid sharp edges.

Use in an environment which meets the following
conditions:●Dry location protected from rain, water, and oils.
●Location not subject to wide temperature changes.
●Location free from cutting chips and dust.
●Horizontal surface which is stable and free from vibration.
●Location protected from use by children and unauthorizedpeople.
・Use in location contrary to the above may cause poor
accuracy, damage to the product, or may result in accident
or injury.

When mounting measurement instrument tobracket clamp, make sure clamp is tightened
firmly.

・If clamp is loose, measurement instrument may falland cause damage or affect instrument accuracy.

①Insure that the measuring surface is free of dirt and is secureand not loose.
Wipe measuring surface with soft cloth moistened with alcohol toremove contamination. Insure that it is tight and is not scratched. Do
not use with damaged or loose platform as it will affect accuracy.
②Check that bracket moves smoothly.
Loosen the bracket clamp and check that it moves smoothly on
the main post. Otherwise, clean post and remove any
contamination or corrosion.
③Attach the measurement instrument to the bracket.
Firmly tighten the bracket clamp.
Loosen the instrument mounting clamp and insert the measuring
instrument stem into the mounting hole. Fix the instrument bytightening the instrument mounting clamp.

Measuring Instrument

２Stem③Fine Adjustmemt Knob
①Bracket④Stem
Mounting Hole

②Bracket Clamp⑤Instrument
Mounting Clamp

３
1

Tighten

Loosen④Taking Measurements.
1. Make sure that there is no dirt or dust on the measuring
platform, measuring instrument, or workpiece.
2. While supporting the bracket by hand, loosen the bracket
clamp and adjust to roughly the desired height before
re-tightening the bracket clamp.
3. Use the fine adjustment knob to fine-tune the height．
Perform the desired measurements.
※Before start of measurement, please check the measuring
surface for wear using the following procedure.

CHECKING THE PLATFORM FOR WEAR
Measuring platform will wear with continued use.
Please check for wear before each use.
①Install an accurate measuring instrument.

②Measure block gauge placed at several points on measuring
surface and make sure the reading is consistent.
If the reading varies significantly, please discontinue use as
the wear on the measuring surface is to large.

Wear and performance degradation will vary
greatly depending on frequency and type of use.
We recommend periodic inspection conforming
your in-house standards and requirements.

３ Tighten

Cleaning and Storage After Use
Wipe measuring platform with a clean, dry cloth or with clothmoistened with alcohol to remove any contamination.

SPECIFICATIONS●Material
Measuring Platform…Granite
Main Post…SteelBracket…Aluminum

Read the instruction manual and follow the instructions.

・Use of product other than as described in the manual
may cause accident.

Insert※Requires 8mm Hex
４
Wrench (Allen Wrench.)※If Bracket clamp is
disassembled, firstassemble the clamp.
①Place the M10 bolt,
spring washer andwasher into the hole in
the Measuring Platform
２as shown in the drawing.
②Mount the Main Post.
③Tighten the bolt tosecure the Main
Post to theMeasuring Platformusing the 8mm HexWrench.
④Slide the Bracket
Clamp onto the Main
Post.
⑤Tighten the Bracket
Clamp to secure in
１position.
8mm Hex Wrench

Always follow these marked operating procedures in order to prevent harm to yourself or others, and to prevent
damage to property.

HOW TO USE

WasherSpring Washer
M10 Bolt②Bracket
Clamp⑤Instrument
Mounting Clamp⑦MeasuringPlatform

SAFETY NOTICEThroughout this manual, this mark:indicates a required condition or procedure.

■PACKAGE CONTENTS ※ Model WGC-150 shown※ Model WGC-150 shown

Please Observe

CAUTION Improper use may result in personal injury or damage to property.

Thank you for purchasing the Niigata Seiki Granite Comparator Stand. This stand holds measuring tools with ȭ8mm stem, including dial gauge and electronic micrometer, for measuring dimensions such as outside diameter,
thickness, and step heights.
● For safe and proper use of this product, please read this instruction manualbefore use and follow the procedures described. Please keep manual where
it is accessible to user for future reference.
● Keep this manual with the instrument if transferred or leased to a third party.
● For inquiries about this product, please contact dealer or place of purchase.

SAFETY PRECAUTIONS

Model

Vertical
Stroke Length

WGC-150135mm

WGC-200250mm

StemBase Flatness Mounting Hole
2μm

ȭ8mm

Fine Adjust Vertical
Stroke Length≧ 1mm

WeightDo not use a stiff brush or abrasive pad as it will damage the surface andcause poor accuracy.

3.0kg

Wipe any contamination from the main post and bracket
using a clean, dry cloth.

6.9kg

Store in area free from dust and with low humidity.

（approx.）Niigata Seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

取扱説明書品 番 WGC-150・WGC-200

精密測定器保持具

石製コンパレータスタンド●第三者に譲渡・貸与される場合も、この説明書を必ず添付してください。

●当商品に関するお問い合わせは、お買い求めの販売店もしくは当社にご連絡ください。

■各部の名称と機能 ※図はWGC-150を使用しています。
③ 微動送りこの説明書は、必ずお守り頂きたい強制事
項にマークを付けて表示しています。
※図はWGC-150を使用しています。

① ブラケット⑥ 主柱① ブラケット

ワッシャ
スプリングワッシャ②ブラケットクランプ⑤ 測定器クランプ⑥ 主柱■組み立て方③微動送り …………… ステム取付穴に差し込んだ測定器の上下位置を微調整します。
④ステム取付穴 ……… 測定器のステムを差し込む取付用の穴です。

⑥主柱 ………………… ブラケットを支える柱です。
⑦測定台 ……………… ワークを置く台です。
使用を繰り返すうちに磨耗しますので、ご使用前に毎回、ブロックゲージを使用して磨耗確認を行ってください。

差し込む⑤測定器クランプ …… ステム取付穴に差し込んだ測定器のステムを、締め付けて固定するクランプです。

取り付け②ブラケットクランプ … ブラケットを締め付けて、主柱に固定するクランプです。

② ブラケット
クランプ

５ 締める

①ワッシャ・スプリング
ワッシャ・M10ボル
トを図 の 順 番で 測
２定台裏の主柱穴へ差し込みます。
②主柱を取り付けます。
③六角棒レンチ8mmでボルトを締め、測定台に主柱を固定してください。
④ブラケットを主柱へ差し込みます。
⑤ブラケットクランプを締めて、ブラケッ
１
トの 位 置を固 定します。
六角棒レンチ8mm

測定作業にのみ使用する。

下記の条件を満たす環境で使用する。

・傷防止のため、ワークは静かに測定台へ載せてください。
また、鋭利なものは載せないでください。

●雨や水、油などがかからない、乾燥した場所●温度変化の少ない場所●切粉や粉塵が発生しない場所●振動やガタツキのない水平な場所●子どもや、使用者以外が近付かない場所
・上記に反する場所での使用は、精度不良や製品の破損、事故やけがの原因となります。

汚れは柔らかい乾いた布か、アルコールを少量含ませた布で拭き取ってください。ガタツキや、測定台上面に傷がある場合は精度が出ないため使用を中止してください。

②ブラケットの動きがスムーズである事を確認するブラケットクランプをゆるめ、ブラケットを上下させます。動
きがスムーズで無い場合は、主柱やブラケットの汚れ、錆などを取り除いてください。

③測定器をブラケットに取り付けるブラケットクランプをしっかりと締めます。
測定器クランプをゆるめて、ステム取付穴に測定器のステムを差し込み、測定器クランプを再度締めて固定してください。

⑥ 主柱

２主柱穴
⑦ 測定台

ワッシャ
スプリングワッシャ
M10ボルト３ 締める

測定台上面の傷に注意する。

測定器を取り付けている時は、ブラケットクランプをしっかりと締めて固定する。

・ブラケットクランプの締め付けがゆるいと、ブラケットごと測定器が落下し、測定器の故障・破損・精度不良の原因となります。

測定器① ブラケット

３ 締める

1. ワークや測定器の測定子にごみ・埃・汚れなどがないことを確認し、ワークを測定台の上に置きます。
2. ブラケットを手で支えながら、ブラケットクランプをゆるめ、おおよその高さに合わせてブラケットクランプを締めます。
3. 微動送りねじで高さを微調整し、任意の位置に合わせたら測定を行ってください。

※測定開始前に、下記手順に従って測定台の磨耗状況の確認を行ってください。

測定台の磨耗状況の確認

測定台は使用を続けると磨耗します。
毎回のご使用前に磨耗状況の確認を行ってください。
①信頼できる測定器を取り付けます。

ステム③ 微動送り④ ステム取付穴⑤ 測定器クランプ④測定する② ブラケット
クランプ②ブロックゲージを測定台の上に置いて数点測定し、お客様の測定に影響するような数値変化が無いか確認してください。
数値変化がお客様の測定に影響する状態になったら、磨耗状況が大きいため使用を中止してください。

性能の劣化は、使用の頻度・環境によって大きく異なります。お客様の社内規格に準じた周期で、定期点検を行うことをおすすめ致します。

1 ゆるめる

使用後のお手入れ・保管方法

測定台の汚れは柔らかい乾いた布か、アルコールを少量含ませた布で拭き取る。

製品仕様

●材質
測定台 …花崗岩
主 柱 …ステンレス
ブラケット…アルミ

本器は大切に取り扱う。

・落下などの衝撃を与えないでください。精度不良や製品の破損の原因となります。
・測定台に水や油が付着した場合は長時間放置せず、柔らかい布などで拭き取ってください。

差し込む①ブラケット ………… 測定器を取り付け、上下に移動します。

４
差し込む※組み立てには、六角棒レンチ8mmが必要です。
※ブラケットクランプが外した状態で梱包されていた場合は、最 初にブラケッ
トクランプをブラケットに取り付けてください。

① ブラケット

・取扱説明書に記載された内容以外での使用は、事故の原因となります。

①測定台に汚れ、傷、ガタツキなどが無いか確認する

M10ボルト

上記の物が揃っているか確認し、部品が足りない場合は、お買い求めの販売店もしくは当社までご連絡ください。

⑦ 測定台

注意 誤った使い方をした場合に『傷害を負う、または物的損害が発生するおそれがある内容』です。

使用方法

⑦ 測定台

④ ステム取付穴

お使いになる人や他の人への危害、財産への損害を未然に防止するため、必ずお守り頂くことを下記のように説明しています。

・指定用途以外へのご使用は、製品の破損や磨耗、予測できない事故の原因となります。

安全上のお知らせ■梱包内容の確認

必ずお守りください。

取扱説明書をよく読み、指示に従う。

この度は「石製コンパレータスタンド」をお買上げ頂きありがとうございます。この商品は、ダイヤルゲージ、電気マイクロメータなど㋡８ステムの測定器を保持し、ワークの外径・厚さ・段差などを測定する測定補助具です。
●正しく安全にお使い頂くため、ご使用の前に本取扱説明書を必ず読み、記載の手順に従ってご使用ください。お読みになった後は、お使いになる方がいつでも見ることができる場所に、大切に保管してください。

安全上のご注意品番

WGC-150

WGC-200

ヘッド上下
有効ストローク

135 mm
250 mm

ベース平面度
2 μm

ステム取付穴㋡ 8 mm

微動送り
上下ストローク

1 mm以上質量

約 3.0 kg

約 6.9 kg

硬いブラシなどで汚れを落とすと測定台に傷がつき、精度不良の原因となるのでおやめください。

主柱やブラケットの汚れを落としてから保管する。

主柱やブラケットは柔らかい布などで汚れを拭き取り、湿度の少ない環境で保管してください。

発売元〒955-0055 新潟県三条市塚野目5丁目3番14号(0256)33-5502(代) FAX(0256)33-5528

URL http://www.niigataseiki.co.jp

I400-K1 2106