DIGITAL BORE GAUGE

Instruction Manual

Model No.：WCDI-10D/ WCDI-18D <combined>

Thank you for purchasing the Digital Bore Gauge.
Please read this manual thoroughly before use for proper operation.

■ PART IDENTIFICATION
Gauge(Digital Indicator)

LCD Display[ABS/SET]
ButtonExtension
RodMeasurement
Head【WCDI-10D】

Length below
Handle:
50mm

Shaft

Probe
Tip

Extension
Rod

Length below Handle:
100mm

HandleMeasurement
Head

Probe
Tip

Guide

【WCDI-18D】【Accessories】（WCDI-10D）（WCDI-18D）（WCDI-10D）（WCDI-18D）
Spanner
Extension Rod※Spanners shown in drawing
Washer 0.5mm（WCDI-18D only）

above vary slightly inappearance, but function will
be same.

Pull out the case from the twoside edges to remove. Insert
battery with (+) side facing
down.
NOTICEUse only CR2032 type lithium
button cell.

■AUTO OFF DELAY

Battery（CR2032）

Battery
Case①Press and hold the [ZERO/ON] Button to turn power OFF.
②Press and hold [ABS/SET ] and [ZERO/ON] Buttons at thesame time.
③When「-----」 is displayed, ❶release the [ZERO/ON] Button,❷then release the [ABS/SET ] Button in that sequence.
④The Auto Off Delay time will be displayed. Press [ABS/SET]
to change.
※Each press of the [ABS/SET ] Button will change the delay
by 1/2 hr. (30 min.) from 0 to 6 hr.
⑤Press [ZERO/ON] when done and display will return to themeasurement screen.
※Once set, the delay time is saved even when power is OFF.

ACCESSORIES●Extension Rods
WCDI-10D ････ 6・6.5・7・7.5・8・8.5・9・9.5・10mm（9 pieces）
WCDI-18D ････ 10・11・12・13・14・15・16・17・18mm（9 pieces）●Washer ･･･････ 0.5mm（WCDI-18D only）●Spanner ･･････ 1x●Battery ･･･････ CR2032 Lithium button cell, 1x（for test）

NOTICE●This is a precision instrument, handle with care.
●Do not disassemble or modify.
●Keep instrument away from water and oil.
●Use only with supplied Extension Rods and Washer.
●Keep instrument away from direct sunlight and high temperatures
such as in a car, or near a stove or heat source.
●After use, remove any dirt or cutting chips and apply rustpreventative oil to Extension Rod and Washer. When not in use
store in supplied storage case.
●This product is for measuring inside diameter; use only as directed.
Improper use may cause accident or injury.
●Avoid using in high electric fields, such as near fluorescent lights orswitching power supplies. Electrical interference may cause
incorrect readings.

■ ASSEMBLY①Loosen the Clamp Screw and insert
the Digital Indicator into the gauge
body. Tighten the Clamp Screw tosecure.

ABS/SETMIN※After attaching, press the [ZERO/ON]
Button on the Digital Indicator to turn onpower. Confirm that the displayed
reading changes by gently squeezing themeasurement head. If necessary, adjust
the depth that the Digital Indicator isinserted into the gauge body.

Ex.）WCDI-18D set to 14.5mm（Fig.3）■ OPERATION①Before use, please make sure the Measurement Head components are not loose and tighten if necessary.
②Clean the Measurement Head and the ID of a Ring Gauge Standard using parts cleaner or cleaning solution.
③Press the [ZERO/ON] Button to turn on power. ※ A long press to the [ZERO/ON] Button will turn power off.

ABSOLUTE MODE (ABS)①Turn on power. If the “※Measure absolute dimension of workpiece” or “” symbols are displayed on the LCD, press the [MIN] Button to turn off.

②Adjust the “SET” value of the Bore Gauge to match the ID of the Ring Gauge Standard. Press and hold the [ABS/SET] Button until“SET” blinks on the display.
③Continue holding the [ABS/SET] Button and the display character to the left will blink. Use a short press of the [ABS/SET] Button tochange the value, and a long press to change the character being modified. Repeat this process to enter the dimension of the Ring
Gauge Standard on the LCD display. (Figure 4)④When the value is entered the flashing position will return to “SET”. A shortpress of the [ABS/SET] Button to turn off the flashing.

⑤Insert the Measurement Head into the Ring Gauge Standard and press the[MIN] Button to put the Indicator into minimum-mode. Slowly pivot thegauge (Figure 3) to display the minimum value.
⑥Remove the gauge from the Ring Gauge and press the [CAL] Button until“OK” is displayed.
Preparation for measurement is now complete.

COMPARATIVE MEASUREMENTS ※Difference between workpiece and reference

Clamp
Screw

NutSpanner

Spanner

Ex. RodWasher
14mm ＋ 0.5mmMIN

OO

ABS/SET

ZERO/ONCALSet to match

Ring Gauge
Standard（Fig.4）

ABSOO

ABS/SETMIN

ZERO/ON※Press the [ON/ZERO] Button momentarily to reset display in preparation
to measure the next workpiece.

■TROUBLESHOOTINGERRORCORRECTIVE ACTION●LCD does not display●Displayed value is unstable●Display is not clear●Cycle power by removing andreplacing battery.
●Replace battery with new one.

Workpiece I.D. value

CAL

Workpiece（Fig.5）②Insert the Measurement Head into the Ring Gauge Standard and press[MIN] Button to put the Indicator into minimum-mode. Slowly pivot thegauge (Figure 3) to display the minimum value.
③Remove the gauge from the Ring Gauge and press the [CAL] Button until“OK” is displayed.
Preparation for measurement is now complete.
④Put the gauge into the workpiece to be measured, and pivot the gauge
slowly. (Figure 3) The LCD will display the difference between themeasured workpiece and the calibration standard. (Figure 6)

Ex. Rod
8.5mm

ABS⑦Put the gauge into the workpiece to be measured, and pivot the gauge
slowly. (Figure 3) The LCD will display the measured value. (Figure 5)※Press the [ON/ZERO] Button momentarily to reset display in preparation
to measure the next workpiece.

①Turn on power. If “” or “” symbols are displayed on the LCD,
press the [MIN] Button to turn off. If “ABS” is displayed press the[ABS/SET] Button to turn off.

Ex.）WCDI-10D set to 8.5mm

Attach the Extension Rod,Washer to the bottom of theMeasurement Head and tighten
with the Spanner.

（Fig.2）

ZERO/ON②Select the Extension Rod, washer (WCDI-18Donly) for the desired
measurement range.
1. Use the Spanner to loosen theNut at the bottom of
Measurement Head.
2. Install the Extension Rod andtighten.

（Fig.1）CAL

INSERT■ INSERTING・REPLACING BATTERY265ｂOO

Cap[CAL] Button※1：Not Used
※2："0.01" indicates the resolution of
the display's digital bar graph.
The bar graph display willchange with each 0.01mm
change in the measured value.

ａZERO/ON

Clamp
Screw

WCDI -18D
10 -180.002
12530.90 〜 6.0300

CAL[MIN]Button

WCDI -10D
6 -100.002
12530 .7

ABS/SET[ZERO/ON]
ButtonZERO/ONCALModel No.
Measuring Range（ｍｍ）
Resolution（ｍｍ）※＊ Wide range accuracy（μｍ）※＊ Adjacent error（μｍ）※＊ Repeatability（μｍ）
Single Stroke Range （ｍｍ）
Auto Off Delay（hｒ）
Weight（ｇ）

ABSOO

ABS/SETMIN※2＊Including supplied Digital Gauge※Not including quantization error (±1 count )MINABS※1■ SPECIFICATIONS■ MEASUREMENT
For accurate measurements, follow the procedure described below.
①In cross section perpendicular to the axis of the bore, as shown in Figure 1, position the gauge along diameter ⓐ. This is themaximum distance, and at this point the Indicator will read the maximum value. The Guide will automatically position the Measurement
Head to diameter ⓐ when inserted.
※WCDI-10D does not have a guide on the Measurement Head so please manually position at this maximum value.
②In cross section parallel to the axis of the bore through diameter ⓐ, position the indicator to minimize the distance ⓑ. (Figure 2)
In this position the Indicator will read the minimum value. This position must be manually determined by pivoting the Gauge as shownin Figure 3 while watching the Indicator to find the minimum value.

OO
ABS/SETMIN※If problem persists, or if you have anyquestions, please contact distributor.
※Please note, manufacturer is unable torespond to inquires or provide service
directly. Please contact distributor.

ZERO/ONCALDiff. between Workpiece and Reference

Workpiece（Fig.6）

Niigata Seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

デジタルシリンダゲージ‫ٯ‬ඏ‑ད࿉‫۽‬ᘹ⬀ nrq p nry a⬂Ԁᘈ⬄

この度はデジタルシリンダゲージをお買上げ頂きありがとうございます。
ご使用前に、本書をよくお読み頂き正しくお使いくださいますようお願い申し上げます。

■各部の名称■仕 様指示器

LCD表示部

CALボタン取付ねじ※1：このボタンは使用しません。
※2：「0.01」表示は液晶表示部外周のバーグラフの単位です。
測定値に0.01mmの変動があった場合それに応じて、バーグラフも1本増減します。

キャップ
にぎり測定部測定子【WCDI-10D】

ご注意

にぎり下長さ100mm

換えロッド

にぎり下長さ50mm

外筒

換えロッド

測定子

測定部

案内板

【WCDI-18D】【付属品】（WCDI-10D）（WCDI-18D）
換えロッド

ワッシャ0.5mm（WCDI-18Dのみ）（WCDI-10D）（WCDI-18D）
スパナ※スパナの形状は上図と若干異なる場合がありますが、使用上問題ありません。

ご注意

電池（CR2032）●精密機器ですので、落としたりぶつけたりしないでください。
●分解・改造はしないでください。
●水や油のかかるところで使用しないでください。
●換えロッド・ワッシャなどは付属のものを使用し、他との併用は避けてください。
●直射日光の当たる場所や炎天下の車の中、火のそばなどに放置しないでください。
●使用後は柔らかい布などでゴミや切粉などを取り除き、換えロッド・ワッシャなどは防錆処理をして、収納ケースの元の位置に保管してください。
●本製品は内径測定用機器です。用途以外のご使用は、事故やけがの原因となりますので絶対におやめください。
●インバータ式蛍光灯などの電界が発生する場所でのご使用は避けてください。LCD表示部の数値が誤作動を起こす原因となります。

■本体の組み付け①取付ねじをゆるめ、指示器を本体に差し込みます。
取付ねじを締めて固定してください。

※固定後、ZERO/ONボタンを押して電源を入れ、測定子を押してLCD表示部の数値が変わることを確認してください。上手く動作しない場
合は、指示器の差込深さを再度調整してください。

電池蓋

電池は必ずCR2032（コイン型リチウム電池）をお使いください。

■オートオフ時間の設定①ZERO/ONボタンを長押しし、電源をOFFにします。
②ABS/SETボタンとZERO/ONボタンを同時に押します。
③LCD表示部に「-----」が表示されたら、❶ZERO/ONボタン→❷ABS/SETボタンの順に放します。
④LCD表示部に設定時間が表示されるので、ABS/SETボタンを押し、希望の時間を表示させます。
※ABS/SETボタンを押すたびに「0.5時間（30分）」ずつオートオフ時間が切り替わります。
（0〜6.0時間）⑤ZERO/ONボタンを押し、画面が測定画面に切り替わったら設定完了です。
※電源をOFFにしても、設定時間は記憶されています。

ABS/SETMIN

ZERO/ON
CAL

取付ねじ②測定する寸法に合わせて、換えロッド・ワッシャ （WCDI-18Dのみ）を選択します。
例）WCDI-10D 8.5mm測定用

１. 付属のスパナで測定部下部のナットをゆるめてください。
２. 換えロッドを差し込み、ナットを締めて固定してください。
例）WCDI-18D 14.5mm測定用付属のスパナで換えロッド・
ワッシャを測定部に固定してください。

換えロッド
8.5mm

スパナ

（図２）

■使用方法

①ご使用の前に、測定部と外筒の間がゆるんでいないか確認し、ゆるんでいたら締め付けてください。
②測定子を含めた測定部と、基準リングゲージの内径をパーツクリーナーなどの洗浄液で洗浄します。
③ZERO/ONボタンを押して、電源をONにしてください。※電源をOFFにする時は、ZERO/ONボタンを長押ししてください。

絶対測定（ＡＢＳ） ※ワークのサイズを絶対値で表示します。
①電源をONにします。LCD表示部に『』、『』のマークが表示されている場合は、MINボタンを押してマークを消してください。

②マスタとなるリングゲージの数値に、シリンダゲージの数値を合わせます。ABS/SETボタンを長押ししてください。SETボタンが点滅します。
③長押しを続けると、点滅位置が変わって行きます。変更したい数値の位置に点滅が移動して来たら、一度ABS/SETボタンを放し、今度は短く押してください。数値が変わります。
これを繰り返し、マスタとなるリングゲージに表示されている実寸法と、LCD表示部の数値を一致させてください。
（図４）④数値が一致したら点滅位置を『SET』まで戻し、ABS/SETボタンを短く押して点滅を解除してください。
⑤測定部をマスタとなるリングゲージに入れ、MINボタンを押して最小値測定モードにします。
（図３）のようにシリンダゲージをゆっくりと振り、最小値を表示させてください。
⑥測定部をマスタとなるリングゲージから抜き、CALボタンを長押ししてください。『OK』が表示されます。
これで測定の準備が完了しました。

ナット
スパナ

ABS

MIN

OO

ABS/SET

ZERO/ON

数値を合わせる

CAL

比較測定※ワークのサイズをマスタとなるリングゲージとの差で表示します。

ABS

OO

ABS/SETMIN

ZERO/ON
CAL

ワークの内径を表示②測定部をマスタとなるリングゲージに入れ、MINボタンを押して最小値測定モードにします。
（図３）のようにシリンダゲージをゆっくりと振り、最小値を表示させてください。

■トラブルと対応
トラブルの内容● 表示が安定しない● 表示が不明確である

対応策● 電池を入れ直してください。
● 新しい電池と交換してください。

ワーク

（図５）

③測定部をマスタとなるリングゲージから抜き、CALボタンを長押ししてください。『OK』が表示されます。
これで測定の準備が完了しました。
④ワークに測定子を入れ、（図３）のようにシリンダゲージをゆっくりと振って、測定してください。LCD表示部に、マスタとなるリングゲー
ジとの差が表示されます。
（図６）※次のワークを測定する際は、ZERO/ONボタンを一度押してください。

マスタとなる
リングゲージ（図４）⑦ワークに測定子を入れ、（図３）のようにシリンダゲージをゆっくりと振って、測定してください。LCD表示部に、測定値が表示されます。
（図５）※次のワークを測定する際は、ZERO/ONボタンを一度押してください。

● ＬＣＤ表示部に何も表示されない

換えロッド
ワッシャ
14mm ＋ 0.5mm（図３）①電源をONにします。LCD表示部に『ABS』、『』、『』のマークが表示されている場合は、ABS/SETボタンとMINボタンを押してマークを消してください。

差し込む■電池のセット・交換
電池蓋両側をツマミながら引き出し、（＋）側が下になるよう電池をセットして、電池蓋を戻してください。

付属品●換えロッドW C D I -10 D･･･ 6・6.5・7・7.5・8・8.5・9・9.5・10mm（計9本）
W C D I -18 D･･･10・11・12・13・14・15・16・17・18mm（計9本）●ワッシャ･･･････ 0.5mm（WCDI-18Dのみ）●ス パナ ････････ 1 本●電 池 ････････ CR2032（コイン型リチウム電池：テスト用）×１個

（図１）

OO

MINボタン

ZERO/ON

ZERO/ON
ボタン

ZERO/ON
CALｂCALOO

ABS/SETMINａ

ABS/SET※1※2

W C D I -10 D
W C D I -18 D
6 -1010 -180 . 0 020 . 0 02121255330.70 .90 〜 6.026 5300

②直径䒏を通る円筒の軸に平行な断面において、直径书は最小値となり、指示器は最小値を表示します。
（図２）
この直径书は、手動で求めます。シリンダゲージを（図３）のように振り、指示器が最小値を示す点を探してください。

ABSABS

品番測 定 範 囲（ｍｍ）
目量（ｍｍ）※＊
広 範 囲 精 度（μｍ）※＊
隣 接 誤 差（μｍ）※＊
繰 返 し 精 密 度（μｍ）
測定子の有効ストローク（ｍｍ）
オートオフ設定（時間）
質量 （ｇ）※ 量子化誤差（±１カウント）含まず①円筒の軸に垂直な断面において、直径䒏は最大値となり、指示器は最大値を表示します。
（図１）
この直径䒏は、可動する案内板によって自動的に求められます。
※WCDI-10Dには案内板がありません。手動で最大値を示す位置を探してください。

MIN

ABS/SET
ボタン（デジタルインジケータ）＊：付属のデジタルインジケータ（指示器）を含めた値です。

■測定方法について正確な数値を測定するために、測定は下記の手順で行ってください。

OO
ABS/SETMIN

ZERO/ONCAL※症状が改善されない場合や不明な点がございましたらお買い上げの販売店、または当社までご連絡ください。
※お問い合わせや、ご連絡が無いまま直接修理品などを送付されても処理、対応ができない場合がありますのでご了承ください。

マスタとなるリング
ゲージとの差を表示

ワーク

（図６）

〒955-0055 新潟県三条市塚野目5丁目3番14号(0256)33-5502(代) FAX(0256)33-5528

URL http://www.niigataseiki.co.jp

2401

I382-K3