Instruction Manual

DIGITAL BORE GAUGE

Model No.：WCDI-6D

Thank you for purchasing the Digital Bore Gauge.
Please read this manual thoroughly before use for proper operation.

■ PART IDENTIFICATION■ SPECIFICATIONS ＊Including supplied Digital Gauge※Not including quantization error (±1 count)

Gauge（Digital Indicator）●Probe Tip ･･････ 4.0 - 4.6mm, 4.5 - 5.1mm
5.0 - 5.6mm, 5.5 - 6.1mm ( 4pcs )●Probe Needle･･･ 1x（※Note: When purchased, Probe Needle[CAL] ButtonCap（Fig. 1）

（Fig. 2）

Probe Tip Probe×４Needle

Probe TipMeasurement
HeadLength
below Handle:42mm

HandleSpanner■ INSERTING・REPLACING BATTERY※1：Not Used※2："0.01" indicates the resolution of
the display's digital bar graph.
The bar graph display willchange with each 0.01mm
change in the measured value.

NOTICE●This is a precision instrument, handle with care.
●Do not disassemble or modify.
●Keep instrument away from water and oil.
●Use only with supplied Probe Tips and Probe Needle.
●Keep instrument away from direct sunlight and high temperatures such asin a car, or near a stove or heat source.
●After use, remove any dirt or cutting chips and apply rust preventative oil
to Probe Tip and Probe Needle. When not in use store in supplied storage
case.
●This product is for measuring inside diameter; use only as directed.
Improper use may cause accident or injury.
●Avoid using in high electric fields, such as near fluorescent lights orswitching power supplies. Electrical interference may cause incorrect
readings.

Pull out the casefrom the two sideedges to remove.
Insert battery with (+)
side facing down.
NOTICEUse only CR2032 typelithium button cell.

Battery（CR2032）■OPERATION

ABS/SETMIN■AUTO OFF DELAY①Press and hold the [ZERO/ON] Button to turnpower OFF.
②Press and hold [ABS/SET ] and [ZERO/ON]
Buttons at the same time.
③When「-----」 is displayed, ❶release the [ZERO/ON]
Button, ❷then release the [ABS/SET ] Button inthat sequence.
④The Auto Off Delay time will be displayed. Press[ABS/SET] to change.
※Each press of the [ABS/SET ] Button will change thedelay by 1/2 hr. (30 min.) from 0 to 6 hr.

⑤Press [ZERO/ON] when done and display will returnto the measurement screen.

※Once set, the delay time is saved even when power is OFF.

ZERO/ON

Probe
NeedleCAL②Insert the Probe Needle into the Probe Tip andscrew onto the end of the Handle. Tighten as
shown using the Spanner to secure. (Figure B,C)

（Fig.Ａ）

Probe
Tip” symbols are displayed on the LCD, press the [MIN] Button to turn off.

②Adjust the “SET” value of the Bore Gauge to match the ID of the Ring Gauge Standard. Press and hold the [ABS/SET] Button until“SET” blinks on the display.
③Continue holding the [ABS/SET] Button and the display character to the left will blink. Use a short press of the [ABS/SET] Button tochange the value, and a long press to change the character being modified. Repeat this process to enter the dimension of the Ring
Gauge Standard on the LCD display. (Figure 4)⑤Insert the Measurement Head into the Ring Gauge Standard and press the[MIN] Button to put the Indicator into minimum-mode. Slowly pivot thegauge (Figure 3) to display the minimum value.
⑥Remove the gauge from the Ring Gauge and press the [CAL] Button until“OK” is displayed.
Preparation for measurement is now complete.

ATTACHCOMPARATIVE MEASUREMENTS ※Difference between workpiece and reference①Turn on power. If “” or “” symbols are displayed on the LCD,
press the [MIN] Button to turn off. If “ABS” is displayed press the[ABS/SET] Button to turn off.

②Insert the Measurement Head into the Ring Gauge Standard and press[MIN] Button to put the Indicator into minimum-mode. Slowly pivot thegauge (Figure 3) to display the minimum value.
③Remove the gauge from the Ring Gauge and press the [CAL] Button until“OK” is displayed.
Preparation for measurement is now complete.

■ TROUBLESHOOTING
ProbeTip（Fig.C）

ExampleABS

mm
ABS/SETMIN

ZERO/ON

Set to match

CALRing Gauge
Standard⑦Put the gauge into the workpiece to be measured, and pivot the gauge
slowly. (Figure 3) The LCD will display the measured value. (Figure 5)※Press the [ON/ZERO] Button momentarily to reset display in preparation
to measure the next workpiece.

※Press the [ON/ZERO] Button momentarily to reset display in preparation
to measure the next workpiece.

Handle（Fig.B）※Measure absolute dimension of workpiece” or “④Put the gauge into the workpiece to be measured, and pivot the gauge
slowly. (Figure 3) The LCD will display the difference between themeasured workpiece and the calibration standard. (Figure 6)

Spanner

INSERT

If necessary, adjust the depth that the Digital
Indicator is inserted into the gauge body.

ClampScrew

INSERT※Select Probe Tip to match the desired
measurement range.
※After attaching, press the [ZERO/ON] Button on
the Digital Indicator to turn on power. Confirm
that the displayed reading changes by gentlysqueezing the measurement head.

ABSOLUTE MODE (ABS)①Turn on power. If the “④When the value is entered the flashing position will return to “SET”. A shortpress of the [ABS/SET] Button to turn off the flashing.

BatteryCase■ ASSEMBLY①Loosen the Clamp Screw and insert the Digital
Indicator into the gauge body. Tighten theClamp Screw to secure. (Figure A)（Fig. 3）

①Before use, please make sure the Measurement Head components are not loose and tighten if necessary.
②Clean the Measurement Head and the ID of a Ring Gauge Standard using parts cleaner or cleaning solution.
③Press the [ZERO/ON] Button to turn on power. ※ A long press to the [ZERO/ON] Button will turn power off.

is inserted into Measurement Head）●Spanner ･･･････ 1x●Battery･････････ CR2032 Lithium cell, 1x (for test)

OO

Clamp Screw

ACCESSORIES

ZERO/ON[ZERO/ON]
Button[MIN]ButtonCAL※1ｂ

ABS/SET[ABS/SET]
ButtonａABS※2

LCD Display

WCDI-6D
4-60.002
10530.6
0 〜 6.0255

MIN

Model No.
Measuring Range （ｍｍ）
Resolution（ｍｍ）※＊ Wide range accuracy （μｍ）※＊ Adjacent error（μｍ）※＊ Repeatability（μｍ）
Single Stroke Range（ｍｍ）
（ｍｍ）
（ hr）
Auto Off DelayWeight（ｇ）■ MEASUREMENT
For accurate measurements, follow the procedure described below.
①In cross section perpendicular to the axis of the bore, as shown in Figure 1, position the gauge along diameter ⓐ. This is themaximum distance, and at this point the Indicator will read the maximum value. The Measurement Head will automatically move todiameter ⓐ when inserted.
②In cross section parallel to the axis of the bore through diameter ⓐ, position the indicator to minimize the distance ⓑ. (Figure 2)
In this position the Indicator will read the minimum value. This position must be manually determined by pivoting the Gauge asshown in Figure 3 while watching the Indicator to find the minimum value.

ERROR

CORRECTIVE ACTION●LCD does not display●Displayed value is unstable●Display is not clear●Cycle power by removing andreplacing battery.
●Replace battery with new one.

（Fig. 4）ABSmm
ABS/SET
MIN

Workpiece

ZERO/ON

Workpiece I.D. value

CAL（Fig. 5）

ABSmm
ABS/SETMIN※If problem persists, or if you have anyquestions, please contact distributor.
※Please note, manufacturer is unable torespond to inquires or provide service
directly. Please contact distributor.

Workpiece

ZERO/ONCAL

Diff. between Workpiece and Reference（Fig. 6）

Niigata Seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp‫ٯ‬ඏ‑ད࿉デジタルシリンダゲージ‫۽‬ᘹ⬀ n⫼⬊

この度はデジタルシリンダゲージをお買上げ頂きありがとうございます。
ご使用前に、本書をよくお読み頂き正しくお使いくださいますようお願い申し上げます。

■各部の名称■仕 様指示器（デジタルインジケータ）

LCD表示部※ 量子化誤差（±１カウント）含まず

目量 （ｍｍ）

WCDI-6D

隣 接 誤 差 （μｍ）

5

繰 返し 精 密 度 （μｍ）

3

測定子の有効ストローク（ｍｍ）

取付ねじ

にぎり

量 （ｇ）

255

測定針

スパナ

※1：このボタンは使用しません。
※2：「0.01」表示は液晶表示部外周のバーグラフの単位です。
測定値に0.01mmの変動があった
場合それに応じて、バーグラフも1本増減します。

5.0-5.6mm、5.5-6.1mm（計４本）● 測定針･･･１本（※購入時、測定針は測定子の中に入っています。）● スパナ･･･１本● 電 池･･･ CR2032（コイン型リチウム電池：テスト用）×１個

電池蓋④数値が一致したら点滅位置を『SET』まで戻し、ABS/SETボタンを短く押して点滅を解除してください。

ご注意⑤測定部をマスタとなるリングゲージに入れ、MINボタンを押して最小値測定モードにします。（図３）のようにシリンダゲージをゆっくりと振り、最小値を表示させてください。

電池は必ずCR2032（コイン型リチウム電池）をお使いください。

測定針

（図Ａ）

測定子

（図Ｂ）

比較測定

※ワークのサイズをマスタとなるリングゲージとの差で表示します。

①電源をONにします。LCD表示部に『ABS』、『』、『』のマークが表示されている場合は、ABS/SETボタンとMINボタンを押してマークを消してください。

（図Ｃ）

ABS

mm
ABS/SETMIN

ZERO/ONCAL

数値を合わせる

マスタとなるリングゲージ

（図４）

ABSmm
ABS/SETMIN

ZERO/ON
CAL

ワークの内径を表示②測定部をマスタとなるリングゲージに入れ、MINボタンを押して最小値測定モードにします。
（図３）のようにシリンダゲージをゆっくりと振り、最小値を表示させてください。

■トラブルと対応
測定子（例）⑦ワークに測定子を入れ、（図３）のようにシリンダゲージをゆっくりと振って、測定してください。LCD表示部に、測定値が表示されます。
（図５）※次のワークを測定する際は、ZERO/ONボタンを一度押してください。

④ワークに測定子を入れ、（図３）のようにシリンダゲージをゆっくりと振って、測定してください。LCD表示部に、マスタとなるリングゲージとの差が表示されます。
（図６）※次のワークを測定する際は、ZERO/ONボタンを一度押してください。

にぎり
取り付ける

⑥測定部をマスタとなるリングゲージから抜き、CALボタンを長押ししてください。『OK』が表示されます。
これで測定の準備が完了しました。

③測定部をマスタとなるリングゲージから抜き、CALボタンを長押ししてください。
『OK』が表示されます。
これで測定の準備が完了しました。

スパナ

差し込む

取付ねじ』のマークが表示されている場合は、MINボタンを押してマークを消してください。

（CR2032）

ZERO/ON差し込む※測定箇所の寸法に合わせて、測定子を付け替えてください。
※本体組付後、ZERO/ONボタンを押して電源を入れ、測定子を押してLCD表示部の数値が変わることを確認してください。上手く動作しない場合は、指示器の差込深さを再度調整してください。

』、『③長押しを続けると、点滅位置が変わって行きます。変更したい数値の位置に点滅が移動して来たら、一度ABS/SETボタンを放し、今度は短く押してください。数値が変わります。
これを繰り返し、マスタとなるリングゲージに表示されている実寸法と、LCD表示部の数値を一致させてください。
（図４）

電池

CAL

②測定子に測定針を差し込み（図Ｂ）、そのままにぎりへねじこみます。
付属のスパナを利用して確実に取り付けてください。
（図Ｃ）①電源をONにします。LCD表示部に『電池蓋両側をツマミながら引き出し、(＋)
側が下になるよう電池をセットして、電池蓋を戻してください。

■本体の組み付け
ABS/SET

絶対測定（ＡＢＳ） ※ワークのサイズを絶対値で表示します。
②マスタとなるリングゲージの数値に、シリンダゲージの数値を合わせます。ABS/SETボタンを長押ししてください。
『SET』マークが点滅します。

①ZERO/ONボタンを長押しし、電源をOFFにします。
● 分解・改造はしないでください。
②ABS/SETボタンとZERO/ONボタンを同時に押します。
● 水や油のかかるところで使用しないでください。
③LCD表示部に「-----」が表示されたら、❶ZERO/ON● 測定子・測定針などは付属のものを使用し、他との併用は避けてください。
ボタン→❷ABS/SETボタンの順に放します。
● 直射日光の当たる場所や炎天下の車の中、火のそばなどに放置しないでく④LCD表示部に設定時間が表示されるので、ABS/SETださい。
ボタンを押し、希望の時間を表示させます。
● 使用後は柔らかい布などでゴミや切粉などを取り除き、測定子・測定針な※ABS/SETボタンを押すたびに「0.5時間（30分）」ずつどは防錆処理をして、収納ケースの元の位置に保管してください。
オートオフ時間が切り替わります。
（0〜6.0時間）● 本製品は内径測定用機器です。用途以外のご使用は、事故やけがの原因と⑤ZERO/ONボタンを押し、画面が測定画面に切り替なりますので絶対におやめください。
わったら設定完了です。
● インバータ式蛍光灯などの電界が発生する場所でのご使用は避けてください。
※電源をOFFにしても、設定時間は記憶されています。
ＬＣＤ表示部の数値が誤作動を起こす原因となります。

MIN③ZERO/ONボタンを押して、電源をONにしてください。※電源をOFFにする時は、ZERO/ONボタンを長押ししてください。

■電池のセット・交換● 精密機器ですので、落としたりぶつけたりしないでください。

①取付ねじをゆるめ、指示器を本体に差し込みます。取付ねじを締めて固定してください。
（図Ａ）■ 使用方法②測定子を含めた測定部と、基準リングゲージの内径をパーツクリーナーなどの洗浄液で洗浄します。

■オートオフの時間設定

ご注意

（図３）

①ご使用の前に、測定部とにぎりの間がゆるんでいないか確認し、ゆるんでいたら締め付けてください。

● 測定子･･･ 4.0-4.6mm、4.5-5.1mm

測定子×４にぎり下長さ42mm

測定部0 〜 6.0

付属品

CALボタンキャップ

測定子質（図２）

（図１）

0.6

オートオフ設定 （時間）ZERO/ONボタン

MINボタンｂmm※＊ａ10ZERO/ON※＊

広 範 囲 精 度 （μｍ）

CAL

ABS/SET
ボタン

4-60.002

ABS/SET※＊ａ を通る円筒の軸に平行な断面において、直径ࠉ
ｂ は最小値となり、指示器は最小値を表示します。
②直径ࠉ
（図２）
ｂ は、手動で求めます。シリンダゲージを（図３）のように振り、指示器が最小値を示す点を探してください。
この直径ࠉ

ABS番ａ は最大値となり、指示器は最大値を表示します。
（図１）①円筒の軸に垂直な断面において、直径ࠉ
ａ は、測定子を差し入れると自動的に求められます。
この直径ࠉ

MIN品

測 定 範 囲 （ｍｍ）

※2

※1＊：付属のデジタルインジケータ（指示器）を含めた値です。

■測定方法について正確な数値を測定するために、測定は下記の手順で行ってください。

トラブルの内容● ＬＣＤ表示部に何も表示されない● 表示が安定しない● 表示が不明確である

対応策● 電池を入れ直してください。
● 新しい電池と交換してください。

ワーク

（図５）

ABSmm
ABS/SETMIN

ZERO/ON
CAL※症状が改善されない場合や不明な点がございましたらお買い上げの販売店、または当社までご連絡ください。
※お問い合わせや、ご連絡が無いまま直接修理品などを送付されても処理、対応ができない場合がありますのでご了承ください。

マスタとなるリング
ゲージとの差を表示

ワーク

（図６）

〒955-0055 新潟県三条市塚野目5丁目3番14号(0256) 33-5502(代) FAX (0256) 33-5528

URL http://www.niigataseiki.co.jp2401

I381-K2