1 
Regulations Governing Verification and Inspection of Measuring Instruments (draft) 
 
[Chronicle of Promulgation and Amendments] 
41 Articles adopted and promulgated by Ministerial Order on 28 August 1987 
Articles 4 and 17 amended and promulgated by Ministerial Order on 14 January 1994 
Articles 4 and 10 amended and promulgated by Ministerial Order on 29 January 1997 
Articles 3, 5, 6, 13, 14, 20, 21, 24, 30, 31, 32, 33, 34, 36, 37, 39 and 40 amended and promulgated; 
and Article 22 deleted by Ministerial Order on 18 August 1999 
Amended and promulgated by Ministerial Order on 23 July 2003 
Articles 3, 22, and 24 amended and promulgated by Ministerial Order on 31 December 2003 
Articles 3 and 5 amended and promulgated by Ministerial Order on 29 December 2004. 
Articles 2, 3, 4 and 6 amended and promulgated by Ministerial Order on 22 December 2006. 
Articles 3,4 and 18 amended and promulgated by Ministerial Order on 22 May 2009. 
Articles 3,12,14 and 22 amended and promulgated by Ministerial Order on 26 March 2010. 
Articles 3,6, and 17 amended and promulgated by Ministerial Order on 31 October 2012. 
Articles 3, 4, 5, 6, 8, 10, 11, 12, 13, 14, 15, 16, 18, 19, 20, 21, and 22 amended and promulgated by 
Ministerial Order on 5 November 2014. 
Articles 3 amended and promulgated by Ministerial Order on 16 November 2015. 
Articles 3, 8, 21 and 22 amended and promulgated by Ministerial Order on 7 May 2018. 
Articles 3 amended and promulgated by Ministerial Order on 16 October 2020. 
Articles 3 amended and promulgated by Ministerial Order on 21 June 2021. 
Articles 3, 12 amended and promulgated by Ministerial Order on 23 May 2022. 
Articles 3, 4, 6, 12 amended and promulgated by Ministerial Order on OO OO 2023. 
 
Article 1  
These Regulations are established in accordance with Article 18 of the Weights and Measures Act 
(hereinafter referred to as “the Act”). 
 
Article 2  
The term “verification” used in these Regulations are classified and defined as follows: 
1. Initial Verification: means the verification of legal measuring instruments subject to verification 
which is conducted before the measuring instruments being released from the production 
premises or imported. However, under the agreement of the dedicated weights and measures 
authority, it can also mean the verification conducted on-site after the measuring instruments 
being installed or while in-use.  
2. Re-verification: means the verification of legal measuring instruments subject to verification 
which is conducted when the validity period of the previous verification expires or after being 
repaired, adjusted or re-modeled, or while in-use. 
The term “verification agency (institution)” used in these Regulations means the dedicated weights 
and measures authority or its branches, or any other government agencies, institutions or 
organizations commissioned by the dedicated weights and measures authority. 
 
Article 3 
The categories and scopes of legal measuring instruments subject to verification are as follows: 
1. Taximeters. 
2. Weighing instruments: non-automatic weighing instruments, automatic gravimetric filling 
weighing instruments and discontinuous totalizing automatic weighing instruments, excluding 
the following instruments:  2(1) The weighing instruments which are not for pricing and transaction use with a number of 
verification scale intervals more than 10 000 and marked that  they are not for transaction use ; 
(2) The weighing instruments with a number of verification scale interval less than or equal to 3000 
and maximum weighing capacity less than or equal to 3  kg marked not for transaction use of 
measuring instruments. 
(3) Portable suspended weighing instruments with a maximum weighing capacity less than or equal 
to 50 kg and marked not for transaction use of measuring instruments. 
(4) Suspended weighing instruments with a maximum weighing capacity more than 1  t; 
(5) Bathroom scales; 
(6) Weighing in motion non-automatic weighing instruments;  
3. Non-Invasive mechanical sphygmomanometers; 
4. Volumeters: 
(1) Liquid volumetric meters: metal measuring pails and measuring tanks marked with divisions; 
excluding the following measuring tanks: 
a. Measuring tanks with a capacity of more than 110 m3. 
b. Pressure measuring tanks. 
(2) Diaphragm gas meters, excluding gas meters with a maximum air flow more than 100 m3/hr. 
(3) Water meters: volumetric water meters, velocity water meters(Woltmann meters, single-jet 
meters and multi-jet meters ）and vortex water meters, excluding combination water meters and 
water meters with a caliber of more than 300 mm. 
(4) Oil meters provided for trading petroleum products, excluding oil meters with a caliber of more 
than 160 mm. 
(5) Liquefied petroleum gas flow meters. 
(6) Air to liquid volumetric ratio (A/L) measuring instruments for official inspection. 
5. Electricity meters ： 
(1)General Electricity meters: Watt-hour meters, Var-hour meters, Watt-hour demand meters, Static 
electricity meters and Instrument transformers, excluding are as follows: 
a. Ancillary electricity meters within the electric products 
b. Ancillary electricity meters within the converters/inverters  
c. Panel meter 
d. Portable electricity meters 
e. Reference electricity meter 
f. Direct current electricity meter 
g. Energy transducer 
h. Standard electricity meters and those with rated voltage higher than 600 V. 
i. Current transformer operated electricity meters those with rated secondary current below 5 A. 
j. Current transformers those with rated secondary current below 5A.   
k. Instrument transformers of 69 kV higher than the nominal system voltage. 
(2)Electric Vehicle Supply Equipment: electric vehicle supply equipment for electric energy 
transaction use, excluding AC electric vehicle supply equipment without electric energy 
measurement. 
6. Speedometers. 
(1) Radar speedometers for law enforcement. 
(2) Laser speedometers for law enforcement. 
(3) Inductive loop speedometers for law enforcement. 
(4) Average speed control devices for law enforcement. 
7. Sound level meters for official inspection. 
8. Concentration meters.  3(1) Breathe alcohol testers and analyzers for official inspection. 
(2) Rice grain moisture meters. 
(3) Field corn moisture meters. 
(4) Vehicles exhaust emissions analyzers for official inspection excluding those used for motorcycles 
and diesel engines. 
9. Illuminance meters for official inspection. 
10. Electrical thermometers. 
The legal measuring instruments subject to verification under the provisions set forth in 
subparagraph 2 may be exempted from verification if they were manufactured or imported before 1 
January 2014 and marked not for transaction use, certification or official inspection. 
The field corn moisture meters listed in item 3, subparagraph 8,Paragraph 1 shall be included as 
legal measuring instruments subject to verification and come into force on 1 July 2016. To meet the 
field corn production season, the period of accepting verification for field corn moisture meters shall 
be during February to March and July to August each year.  
The average speed control devices for law enforcement listed in item 4, subparagraph 6, Paragraph 
1 shall be included as legal measuring instruments subject to verification and enter into force on 1 
January 2021. 
The air to liquid volumetric ratio (A/L) measuring instruments for official inspection listed in item 6, 
subparagraph 4, Paragraph 1 shall be included as legal measuring instruments subject to verification 
and enter into force on 1 October 2021. 
Electric vehicle supply equipment listed in item 2, subparagraph 5, Paragraph 1 shall be included as 
legal measuring instruments subject to verification and enter into force on 1 January 2023. However, 
before the date of implementation, electric vehicle supply equipment that have been released out of 
the factory or imported will be subject to verification on 1 January 2025. 
The applicable targets, applicable legislations, and usage of the legal measuring instruments for 
official inspection or law enforcement in subparagraph 4,6 to 9, Paragraph 1 are shown in Annex. 
The definitions of petroleum products used in item 4, subparagraph 4, Paragraph 1 shall be referred 
to subparagraph 4, Paragraph 1, Article 2 of Petroleum Administration Act and include gasoline, 
diesel oil, kerosene, naphtha, jet fuel, and fuel oil. 
Article 4 
The markings as referred to in item 1 to 3, subparagraph 2, Paragraph 1, Article 3 shall be marked 
not for transaction use in Chinese on the front of the body of measuring instruments and on the 
obvious place of the package or container. 
The markings on the body of measuring instruments referred to in the preceding Paragraph shall be 
carved or printed on the body or made in other permanent manner. 
Where markings are not made in accordance with the provisions of the preceding two paragraphs, it 
shall be ordered to rectify the unlawful act within a given time limit, and if such unlawful act is not 
rectified upon expiry of the deadline date, verification shall be conducted in accordance with 
relevant provisions. 
Article 5 
The applicant for initial verification shall be the manufacturer or importer of the legal measuring 
instruments subject to verification. Under special circumstances with the agreement of the 
verification agency (institution), the applicant may be the owner or holder of the legal measuring 
instruments. 
Once the type approval of legal measuring instruments subject to verification is transferred to any 
other individuals or companies, the original applicant no longer has the right to apply for initial  4verification of the same type of the legal measuring instruments. 
 
Article 6 
To apply for verification, the applicant shall prepare an application form and submit it to the 
verification agency (institution) together with the verification fee, the legal measuring instruments 
subject to verification and the required documents specified in the technical specifications for 
verification and inspection of measuring instruments. Where necessary, the verification agency 
(institution) may notify the applicant to provide other relevant documents. 
The application form as referred to in the preceding Paragraph shall include the following contents: 
1. Name, address and signature of the applicant  (Signature may be exempted for online 
applications);  
2. License number of the weights and measures business license. However, this is not applicable if 
the applicant is not a weights and measures business operator; and 
3. Category, quantity and content of the legal measuring instruments subject to verification. 
 
Article 7 
Upon receiving an application for verification, the verification agency (institution) shall notify the 
applicant to rectify within thirty days if the application is found not to conform with the 
requirements after document review. If the applicant fails to rectify by the expiration of the 
preceding period, the verification agency (institution) shall reject the application and return the 
verification fee. 
 
Article 8 
Verification shall be conducted on the premises of the verification agency (institution). Under 
special circumstances, the applicant may apply to the verification agency (institution) for 
dispatching its personnel to conduct verification on the site where the legal measuring instruments 
subject to verification are installed. 
Where the verification is conducted on-site, the verification working standards used to verify the 
legal measuring instruments shall be carried to the site with the personnel of the verification agency 
(institution), or be the ones which are assessed and found to conform by the verification agency 
(institution). 
In case the personnel of the verification agency (institution) carries the verification working 
standards with them to conduct verification on-site as mentioned in the preceding Paragraph, the 
verification agency (institution) may seek the assistance from the applicant to transport the 
verification working standards to the site. 
 
Article 9 
Where the content, quantity or marking of the measuring instruments are found inconsistent with the 
description in the application form and the applicant cannot rectify in time, the verification agency 
(institution) shall not conduct the verification. 
The applicant shall make rectifications as referred to in the preceding Paragraph within twenty days. 
If the applicant fails to rectify by the expiration of the preceding period, the application shall be 
rejected. If the inconsistencies involve deceitful behavior, the verification agency (institution) shall 
reject the application. 
  5Article 10 
Where the personnel is dispatched to conduct the verification on-site, if the verification can not be 
done due to causes not attributable to the applicant, the verification agency (institution) may 
designate another date for conducting verification, and the applicant need not to pay the verification 
fee again. 
If the verification cannot be conducted due to causes attributable to the applicant, the result of 
verification shall be deemed as incompliance. 
 
Article 11 
The division value of the scale of a legal measuring instrument subject to verification shall employ 
“1,” “2” or “5” multiplied by the power of “10,” and the power shall be positive or negative integer 
or zero. 
 
Article 12 
The verification shall be conducted one by one for all legal measuring instruments subject to 
verification. However, the verification of weighing instruments listed in subparagraph 2,Paragraph 1, 
Article 3 and electrical thermometers listed in subparagraph 10 may be conducted on the basis of 
sampling. 
The implementation of sampling verification referred to in the preceding paragraph shall be 
stipulated by technical specification for the verification and inspection of respective legal measuring 
instruments. 
Where necessary, the measuring instruments may be disassembled to conduct the verification 
referred to in the first Paragraph. The same practice shall apply for inspection. 
 
Article 13 
The record of verification of legal measuring instruments subject to verification shall be made and 
include the following items: 
1. Name of the applicant; 
2. Category, trademark, model and serial number of the legal measuring instruments subject to 
verification and other relevant data; 
3. Maximum permissible error, measurement error, results of verification (compliance or 
incompliance) and the number of verification certificate of the legal measuring instruments 
subject to verification; 
4. Category and serial number of the measurement standard and the number of the verification 
record; 
5. The site where the verification is conducted and the name of personnel conducting verification; 
and 
6. Date of verification. 
 
Article 14 
Legal measuring instruments that have passed verification shall bear a verification mark “
 ,” the 
ratio of the lines to the spaces being 1:2. 
The mark “
 ” shall be affixed to the measuring instruments in any irremovable methods of the 
following manners or in their combination: 
1. Engraving, stamping, spraying, branding or adhering.  62. Seal. 
The verification agency (institution) shall issue a report of verification result of legal measuring 
instruments after conducting verification. 
For legal measuring instruments that meet the requirements of the technical specifications for 
verification and inspection of measuring instruments, a verification certificate shall be issued after 
they pass verification. An applicant may also apply for issuance of verification certificates by 
describing the reason and paying the certificate fee. 
 
Article 15 
The verification records of legal measuring instruments that have passed verification shall be 
maintained for two years in addition to the validity period of the verification. Where the validity 
period of the verification is not prescribed, the records shall be kept for fifteen years. 
 
Article 16 
An application for reissue of the original verification certificate may be submitted to the verification 
agency (institution) by submitting an application form (or relevant supporting documents) and 
certificate fee if the original certificate is lost or damaged. 
 
Article 17 
The legal measuring instruments that have passed verification shall be subject to periodical or 
non-periodical inspection conducted by the verification agency (institution) when they are put in 
use. 
The verification agency (institution) may decide the scope of inspection according to all or, part of 
verification and inspection technical requirements of legal measuring instruments. 
 
Article 18 
Where a verified legal measuring instrument is suspended from use due to certain causes, the owner 
or holder may report to the verification agency (institution), who shall attach slips of suspension of 
use to the legal measuring instrument, and the legal measuring instrument may be exempted from 
inspection. 
Where the suspension of use referred to in the preceding Paragraph is not reported to the verification 
agency (institution) and the validity period of verification has expired, the owner or holder shall 
make new applications for verification. 
 
Article 19 
Inspection shall be conducted on the site where the legal measuring instruments are put in use 
during the business hours of the organizations under inspection. If the inspection cannot be 
conducted on the site of usage, the verification agency (institution) may notify the owner or holder 
to send the legal measuring instruments to a designated location for inspection within a given time 
limit. 
 
Article 20 
The legal measuring instruments that have passed inspection shall bear the inspection compliance  7mark on their front bodies respectively. However, taximeters may bear the inspection compliance 
mark on other place specified by the verification agency (institution). 
The pattern of the inspection compliance mark referred to in the preceding Paragraph shall be “
 ,” 
the ratio of the lines to the spaces being 1:1. 
 
Article 21 
Under any of the following circumstances, the verification agency (institution) shall remove the 
verification and inspection compliance marks (including the corresponding seals) from the verified 
legal measuring instruments, but in case of multi-instruments sharing a seal and at least one of them 
passed inspection or verification the seal shall be kept: 
1. Where they do not pass re-verification; or 
2. Where they do not pass inspection. 
The measuring instruments that do not pass inspection as referred in the preceding Paragraph shall 
be attached with markings of stopping use. 
 
Article 22 
The formats of the verification and inspection compliance marks and verification certificates shall 
be prescribed by the dedicated weights and measures authority. 
 
Article 23 
These Regulations shall come into force upon the date of promulgation unless another date of 
enforcement is prescribed. 
 
 
Note: In case of any discrepancy between this translation and the Chinese version, the Chinese 
version shall govern 
  8Annex   
The applicable targets, applicable legislations, and usage of the legal measuring instruments for 
official inspection or law enforcement 
Legal measuring instruments for 
official inspection or law enforcement  Scope 
Radar speedometers for law 
enforcement Used by police authorities for reporting violations 
in accordance with the Road Traffic Management 
and Penalty Act. 
Laser speedometers for law 
enforcement 1. Used by police authorities for reporting 
violations in accordance with the Road Traffic 
Management and Penalty Act; 
2. Used by special municipalities or counties 
(cities) for reporting  violations of the boats’ 
speed limits in the water area in accordance 
with the self-government ordinances of special 
municipalities or counties (cities); and 
3. Used by reservoir authorities for reporting 
violations in accordance with the Water Act. 
Inductive loop speedometers for law 
enforcement Used by police authorities for reporting violations 
in accordance with the Road Traffic Management 
and Penalty Act. 
Average speed control devices for law 
enforcement Used by police authorities for reporting violations 
in accordance with the Road Traffic Management 
and Penalty Act. 
Sound level meters for official 
inspection. 1. Used by environmental protection authorities 
for imposing penalties and by approved noise 
analysis laboratories for performing inspection 
and testing in accordance with the Noise 
Control Act; 
2. Used by authorities of civil and military 
airports for performing inspection and testing 
in accordance with Noise Control Act; and  
3. Used by labor inspection authorities for 
implementing the Occupational Safety and 
Health Act.  
Breath alcohol testers and analyzers for 
official inspection 1. Used by police authorities for implementing 
Article 185-3, Criminal Code of the Republic 
of China or the Road Traffic Management and 
Penalty Act; and 
2. Used by civil aeronautics authorities for  9implementing Civil Aviation Act. 
Vehicles exhaust emissions analyzers 
for official inspection excluding those 
used for motorcycles and diesel engines  1. Used by analysis laboratories which are 
approved by the central competent authority for 
performing air pollution monitoring in 
accordance with the Air Pollution Control Act; 
and 
2. Used by motor vehicle authorities for 
implementing the Air Pollution Control Act. 
Air to liquid volumetric ratio (A/L) 
measuring instruments for official 
inspection Used by environmental protection authorities for 
imposing penalties and by approved analysis 
laboratories for performing air pollution monitoring 
in accordance with the Air Pollution Control Act.. 
Illuminance meters for official 
inspection 1. Used by labor inspection authorities for 
implementing the Occupational Safety and 
Health Act; and 
2. Used by environmental protection authorities 
for imposing penalties and by approved 
analysis laboratories for performing air 
pollution monitoring in accordance with the 
Air Pollution Control Act.