1ICS19.080；71.040.10
CCSN20；N61
中华人民共和国国家标准
GB4793-××××
测量、控制和实验室用电气设备安全
技术规范
Safetyspecificationforelectricalequipmentformeasurement,controlandlaboratoryuse
（征求意见稿）
在提交反馈意见时，请将您知道的相关专利连同支持性文件一并附上。
××××-××-××发布 ××××-××-××实施
GB4793-××××
I目次
前言..............................................................................IV
1范围.................................................................................1
2规范性引用文件.......................................................................1
3术语和定义...........................................................................1
4环境条件.............................................................................1
4.1正常的环境条件.....................................................................1
4.2扩展的环境条件.....................................................................1
5标志和文件...........................................................................2
5.1标志...............................................................................2
5.2警告标志...........................................................................4
5.3标志耐久性.........................................................................4
5.4文件...............................................................................4
6防电击...............................................................................6
6.1概述...............................................................................6
6.2例外...............................................................................7
6.3危险带电的限值.....................................................................7
6.4基本防护措施.......................................................................8
6.5附加防护措施.......................................................................8
6.6绝缘要求..........................................................................10
7防机械危险..........................................................................17
7.1概述..............................................................................17
7.2锋利边缘..........................................................................18
7.3运动零部件........................................................................18
7.4稳定性............................................................................21
7.5提起和搬运用装置..................................................................21
7.6墙壁安装..........................................................................21
7.7飞散的零部件......................................................................21
8耐机械应力..........................................................................21
9防止火焰蔓延........................................................................21
9.1概述..............................................................................21
9.2消除或减少设备的引燃源............................................................22
9.3一旦出现着火，将火焰控制在设备内..................................................23
9.4限能电路..........................................................................24
9.5对装有或使用可燃液体设备的要求....................................................25
9.6过流保护..........................................................................26
10设备的温度限值和耐热...............................................................26
10.1对防灼伤的表面温度限值...........................................................26
10.2绕组的温度.......................................................................26
10.3耐热.............................................................................27GB4793-××××
II11防止流体和固体异物危险.............................................................27
11.1概述.............................................................................27
11.2清洗.............................................................................27
11.3洒落.............................................................................27
11.4溢出.............................................................................27
11.5电池电解液.......................................................................27
11.6具有额定防护等级的设备（IP代码）.................................................27
11.7流体压力和泄漏...................................................................27
12防辐射（包括激光源）、声压和超声压.................................................28
12.1概述.............................................................................28
12.2产生电离辐射的设备...............................................................28
12.3光辐射...........................................................................29
12.4微波辐射.........................................................................29
12.5声压和超声压.....................................................................29
12.6激光源...........................................................................29
13对释放的气体和物质、爆炸和内爆的防护...............................................29
13.1有毒和有害气体和物质.............................................................29
13.2爆炸和内爆.......................................................................29
14元器件和组件.......................................................................30
14.1概述.............................................................................30
14.2电动机...........................................................................30
14.3过温保护装置.....................................................................30
14.4熔断器座.........................................................................30
14.5电网电源电压选择装置.............................................................30
14.6印制线路板.......................................................................30
14.7用于限制瞬态过电压的电路.........................................................30
15利用联锁装置的保护.................................................................30
15.1概述.............................................................................30
15.2防止重新启动.....................................................................31
15.3可靠性...........................................................................31
附录A（规范性）6.6中未涵盖的绝缘要求...............................................32
A.1电网电源电路的绝缘................................................................32
A.1.1概述............................................................................32
A.1.2电网电源电路电气间隙和爬电距离..................................................32
A.1.3电网电源电路的固体绝缘..........................................................34
A.2二次电路中的绝缘..................................................................37
A.2.1概述.............................................................................37
A.2.2电气间隙........................................................................37
A.2.3爬电距离........................................................................40
A.2.4固体绝缘........................................................................41
A.3条款6.7、A.1或条款A.2中未涉及的电路绝缘.........................................43
A.3.1概述............................................................................43
A.3.2电气间隙计算....................................................................43
A.3.4爬电距离........................................................................46GB4793-××××
IIIA.3.5固体绝缘........................................................................46
参考文献..............................................................................48
图1电击防护措施可接受的方案...........................................................6
图2正常条件和单一故障条件下电容限值对应电压...........................................8
图3连接螺钉组件示例..................................................................10
图4两层间界面上的导体的间距..........................................................12
图5沿着两个内层界面的相邻导体的间距..................................................13
图6相同层之间的相邻导体的间距........................................................13
图4防止火焰蔓延要求的流程图..........................................................22
图6结构符合9.3.2c)1)规定的外壳底部的区域..........................................24
表1符号...............................................................................2
表2海拔5000m内的电气间隙倍增系数...................................................11
表3不超过300V的过电压类别Ⅱ的电网电源电路的电气间隙和爬电距离......................11
表4不超过300V的过电压类别Ⅱ的电网电源电路的固体绝缘的试验电压......................13
表5不超过300V的过电压类别Ⅱ的由电网电源电路供电的二次电路的电气间隙和试验电压......15
表6二次电路的爬电距离................................................................15
表7距离或厚度最小值(见6.6.3.4.2～6.6.3.4.4).......................................17
表2对身体部位构成机械危险的防护措施..................................................19
表3防止不同身体部位挤压的最小保持间隙................................................19
表4防止不同身体部位进入的最大间隙....................................................20
表5外壳底部允许的开孔................................................................23
表6最大可获得电流值的限值...........................................................25
表7过流保护装置的值..................................................................25
表8正常条件下表面温度限值............................................................26
表9绕组绝缘材料的最高温度............................................................27GB4793-××××
IV前言
GB4793的全部技术内容为强制性。
本文件按照GB/T1.1-2020《标准化工作导则第1部分：标准化文件的结构和起草规则》的规定
起草。
本文件代替以下国家标准：
——GB4793.1-2007《测量、控制和实验室用电气设备的安全要求第1部分：通用要求》；
——GB4793.2-2008《测量、控制和实验室用电气设备的安全要求第2部分：电工测量和试验用
手持和手操电流传感器的特殊要求》；
——GB4793.3-2008《测量、控制和实验室用电气设备的安全要求第3部分：实验室用混合和搅
拌设备的特殊要求》；
——GB4793.5-2008《测量、控制和实验室用电气设备的安全要求第5部分：电工测量和试验用
手持探头组件的安全要求》；
——GB4793.6-2008《测量、控制和实验室用电气设备的安全要求第6部分：实验室用材料加热
设备的特殊要求》；
——GB4793.7-2008《测量、控制和实验室用电气设备的安全要求第7部分：实验室用离心机的
特殊要求》；
——GB4793.9-2013《测量、控制和实验室用电气设备的安全要求第9部分：实验室用分析和其
他目的自动和半自动设备的特殊要求》。
本文件主要技术变化如下：
a)从GB4793系列标准中提取基本要求，制定本文件；
b)本文件的技术内容只规定了技术要求，不给出对应的试验方法。
GB4793.1-2007对应的国际标准IEC61010-1的版本发生了变化，最新版本为IEC61010-1:2017。
因此本文件编制过程中，以IEC61010-1:2017为主要参考依据。
本文件与GB/T42125系列标准配套使用。GB/T42125系列标准包含了本文件技术要求对应的试验
方法。
请注意本文件的某些内容可能涉及专利。本文件的发布机构不承担识别专利的责任。
本文件由工业和信息化部提出。
本文件由工业和信息化部归口。
本文件的起草单位：
本文件的主要起草人：GB4793-××××
1测量、控制和实验室用电气设备安全技术规范
1范围
本文件规定了测量、控制和实验室用电气设备的标志和文件、防电击、防机械危险、耐机械应力、
防止火焰蔓延、设备的温度限值和耐热、防止流体和固体异物的危险、防辐射（包括激光源）和声压以
及超声压、对释放的气体和物质以及爆炸和内爆的防护、元器件和组件、利用联锁装置的保护等安全要
求。
本文件适用于测量、控制和实验室用电气设备，包括工业自动化设备。
2规范性引用文件
下列文件中的内容通过文中的规范性引用而构成本文件必不可少的条款。其中，注日期的引用文件，
仅该日期对应的版本适用于本文件；不注日期的引用文件，其最新版本（包括所有的修改单）适用于本
文件。
GB/T4025人机界面标志标识的基本和安全规则指示器和操作器件的编码规则
GB/T4208外壳防护等级(IP代码)
GB/T5169.16电工电子产品着火危险试验第16部分：试验火焰50W水平与垂直火焰试验方法
GB7247.1激光产品的安全第1部分：设备分类、要求和用户指南
GB/T19661(所有部分)核仪器及系统安全要求
GB/T20138电器设备外壳对外界机械碰撞的防护等级（IK代码）
IEC60027电工用文字符号
3术语和定义
GB/T42125.1界定的术语和定义适用于本文件。
4环境条件
4.1正常的环境条件
本文件适用于被设计成至少在下述条件下使用是安全的设备：
a)室内使用；
b)海拔高度不超过2000m；
c)温度在5℃～40℃；
d)温度低于31℃时最大相对湿度为80%，线性降到温度为40℃时相对湿度50%；
e)电网电源电压波动不超过标称电压的±10%；
f)瞬态过电压达到过电压类别Ⅱ的等级；
注1：瞬态过电压的这些等级是由建筑物电线线路供电的设备的典型值。
g)发生在电网电源上的暂态过电压；
h)预定环境适用的污染等级（大多数情况下是污染等级2）。
注2：制造商可以规定更加受限制的操作环境条件，但是设备在这些正常的环境条件下必须是安全的。
4.2扩展的环境条件
本文件适用于被设计成不仅在4.1规定的环境条件下使用是安全的设备，而且在设备制造商规定的
下列额定条件下使用仍是安全的设备。GB4793-××××
2a)室外使用；
b)海拔高度超过2000m；
c)环境温度低于5℃或高于40℃；
d)相对湿度高于4.1的规定；
e)电网电源的电压波动超过标称电压的±10%；
f)潮湿场所；
g)瞬态过电压达到过电压类别Ⅲ或Ⅳ的等级。
5标志和文件
5.1标志
5.1.1概述
设备上应标有符合5.1.2～5.2规定的标志。除了内部零部件的标志外，这些标志应从外部就能看
见，或者，如果盖子或门是预定要由操作人员来拆下或打开的，则在不用工具拆下盖子或打开门后，这
些标志应从外部就能看见。适用于整台设备的标志不应标在操作人员不用工具就能拆卸的零部件上。
对机柜安装或面板安装的设备，标志允许标在设备从机柜或面板上卸下之后能看见的表面上。
量值和单位的文字符号应符合IEC60027的规定，如果适用，图形符号应符合表1的规定。图形符
号应在文件中进行解释。
5.1.2标识
设备应至少标有下列内容：
a)制造商或供应商的名称或商标；
b)型号、名称或能识别设备的其他方法。如果标有相同识别标志（型号）的设备是在一个以上的
生产场地制造的，则对每一个生产场地制造的设备，其标志应能识别出设备的生产场地。
5.1.3电源
设备应标有以下信息：
a)电源性质：
1)交流：额定电网电源频率或频率范围；
2)直流：表1序号1的符号。
b)额定电源电压或额定电源电压范围；
c)连接上所有附件或插件模块时的最大额定功率，单位为瓦（W）（有功功率）或单位为伏安（VA）
（视在功率），或者最大额定输入电流。如果设备可以使用一个以上的电压范围，则应对应每
个电压范围分别标出，除非最大值与最小值相差不大于平均值的20%。标志值不应小于最大值
的90%。
d)对操作人员能设置成使用不同额定电源电压的设备，应装有设置设备电压的指示装置。对于便
携式设备，该电压指示应从外部就能看见。如果设备在结构上做成不用工具就能改变电源电压
的设置，则在改变电压设置的操作时也应能同时改变电压的指示。
e)对能插入标准电源插头的辅助电源插座，如果其供电与电网电源电压不同，则应标出该供电电
压。如果该插座仅供特定的设备使用，则该插座的标志应能识别预定与其使用的设备，如果不
标这种标志，则应标出最大额定电流或功率，或者在插座旁标上表1序号13的符号，并将全
部细节在文件中作出说明。
表1符号
序号 符号 标准 说明
1
 GB/T5465.2-5031(2002-10) 直流
2
 GB/T5465.2-5032(2002-10) 交流
3
 GB/T5465.2-5033(2002-10) 交直流GB4793-××××
34
 GB/T5465.2-5032(2002-10) 三相交流
5
 GB/T5465.2-5017(2006-08) 地（大地）端子
6
 GB/T5465.2-5019(2006-08) 保护导体端子
7
 GB/T5465.2-5020(2002-10) 机箱或机架端子
8
 GB/T5465.2-5007(2009-02) 通(电源)
9
 GB/T5465.2-5008(2009-02) 断(电源)
10
 GB/T5465.2-5172(2003-02)全部由双重绝缘或
加强绝缘保护的设
备
11
 小心，电击危险
12
 GB/T5465.2-5041(2002-10)小心，热表面
13
 ISO7000-0434B(2004-01)小心
14
 GB/T5465.2-5268(2002-10)双位按钮控制的“按
入”状态
15
 GB/T5465.2-5269(2002-10)双位按钮控制的“弹
出”状态
16
 ISO361 电离辐射
5.1.4熔断器
对可由操作人员更换的任何熔断器应在其熔断器座旁标上使操作人员能识别正确更换熔断器的标
志。
5.1.5端子、连接件和操作装置
5.1.5.1概述
如果对安全有必要，应对端子、连接器、控制件以及指示器，包括供流体，如气体、水和供排放用
的任何连接件应给出其用途的指示。如果没有足够的空间，可以使用表1序号13的符号。
急停装置的按钮和执行器以及仅用于指示危险警告或需要紧急行动的指示器应标为红色，并按照
GB/T4025的规定编码。如果颜色的意义与人员或环境的安全相关，则应提供附加的编码方法（见GB/T
4025）。
5.1.5.2端子
与电网电源相连的端子应是能识别的。下列端子应按下面规定进行标志：
a)功能接地端子用表1序号5的符号；
b)保护导体端子用表1序号6的符号，但当保护导体端子是经认可的电网电源器具输入插座的一
部分时除外。该符号应标在靠近端子处或标在端子上；
c)对允许与可触及导电零部件相连的控制电路端子，如果该端子的这种连接不是显而易见的，则
用表1序号7的符号；GB4793-××××
4d)从设备内部获得供电的而且是危险带电的端子应标上电压、电流、电荷或能量的值或量程，或
者标上表1序号13的符号。本要求不适用于使用标准电源插座的电源插座。
5.1.6开关和断路器
如果电源开关或断路器被用来作为断开装置，则应清楚地标出其“断”位。
在某些情况下，表1序号8和9的符号也能适合作为该装置的标识。仅有指示灯不认为是符合要求
的标志。
如果按钮开关被用来作为电源开关，则可以用表1序号8和序号14的符号来表示“通”位，或可
以用表1序号9和序号15的符号来表示“断”位，并将这一对符号（序号8和序号14，或序号9和序
号15）靠近在一起。
5.1.7用双重绝缘或加强绝缘保护的设备
只有局部用双重绝缘或加强绝缘保护的设备不应标上表1序号10的符号。
5.1.8现场接线端子盒
正常条件下，在环境温度为40℃时，或在最高额定的环境温度（如果高于40℃时）下，如果现场
接线盒或接线箱的端子或外壳的温度超过60℃，则应有标志警示安装者在决定连接到端子电缆的温度
额定值时要参考安装说明书。该标志应在连接前或连接时就能看见，或者将该标志标在端子的近旁。表
1序号13的符号是可以接受的标志。
5.2警告标志
警告标志在设备准备作正常使用时就能看见。如果某个警告标志适用于设备的某个特定部分，则该
标志应标在该特定部分上或标在其附近。
警告标志的尺寸应按如下规定：
a)符号高度至少应为2.75mm，文字高度至少应为1.5mm，文字在颜色上应与背景颜色形成反差。
b)在材料上模注、模压或蚀刻的符号或文字的高度至少应为2.0mm。如果不打算在颜色上形成
反差，则这些符号或文字至少应具有0.5mm的凹陷深度或凸起高度。
如果为了保持设备提供的防护而需要责任者或操作人员去查阅说明书，则设备应标有表1序号13
的符号。如果已经使用了其他针对安全方面的符号，且在文件中有解释，则不要求使用表1序号13的
符号。
如果使用说明，允许操作者使用工具触及在正常使用中可能会造成危险的零件或位置，则应有警告
标志，指示在触及之前必须将设备置于安全状态。
5.3标志耐久性
要求的标志应在正常使用条件下保持清晰可辨，并能耐受由制造商规定的清洁剂的影响。
5.4文件
5.4.1概述
出于安全目的，基于操作人员和责任者的需要，应将下述文件随设备一起提供。应保证由制造商授
权的服务人员能获得他们所需要的安全文件：
a)设备的预定用途；
b)技术规范；
c)可从其获得技术支持的制造商或供货商的名称和地址；
d)5.4.2～5.4.6规定的信息；
e)对于出于安全原因要求使用具有规定特性的专用附件（如探头组件）的设备，文件中应指出只
能使用满足制造商规格要求的附件；
f)如测量、指示或检测危险性或腐蚀性物质时的错误读数或危险带电电量可能导致危险时，则说
明书应提供有关如何确认设备功能正常的指南；
g)提起和搬运说明。
标志在设备上的警告符号和警告语句应在文件中给出解释。特别是应给出一段叙述，说明在标有表GB4793-××××
51序号13的符号的所有的情况下均需要查阅文件，以便弄清潜在危险的性质以及必须采取的任何应对
措施。
提供的文件可以是纸质的或电子的，但是当安全必需的所有信息在需要时可能不能通过电子形式获
取时，要求必须有纸质文件。文件应与设备一起交付。应考虑到责任者阅读文件的能力。
5.4.2设备额定
文件应包含以下内容：
a)电源电压或电压范围，频率或频率范围，以及功率或电流额定值；
b)所有输入和输出连接的说明；
c)为设备设计给定的环境条件范围的说明；
d)如果设备有符合GB/T4208的标识，设备防护等级（IP）的说明。
5.4.3设备安装
文件应包括安装和特定的调试说明，以及如果对安全是必要的话，还应包括在设备安装和调试过程
中或由于设备安装或调试不当而引起的危险警告。此类信息（如果适用）包括：
a)装配、定位和安装要求；
b)保护接地说明；
c)与电源的连接；
d)对永久性连接式设备：
1)电源布线要求；
2)对任何外部开关或断路器和外部过流保护装置的要求，以及将这些开关或电路断路器设置
在设备近旁的建议；
e)通风要求；
f)特殊外部服务的要求和安全特性；
g)与声级有关的说明。
5.4.4设备操作
如果适用，使用说明应包括：
a)操作控制件及其用于各种操作方式的标识和描述；
b)不要将设备放在难以操作断开装置的位置的说明；
c)与附件和其他设备互连的说明，包括指出适用的附件、可拆卸的零部件和任何专用的材料；
d)间歇工作限值的规范；
e)在设备上使用的与安全有关的符号的解释；
f)消耗材料更换的说明；
g)清洗和消毒的说明；
h)列出设备中能释放的任何潜在的有毒或有害的物质及其可能的释放量的说明；
i)关于可燃液体风险降低的程序的详细说明；
j)降低来自允许超过8.1温度限值表面的灼伤风险的方法细节。
说明书中应声明，如果不按制造商规定的方法来使用设备，则可能会损害设备所提供的防护。
5.4.5设备维护和维修
应向责任者提供足够详细的说明，以确保对设备进行安全的维护、检查和测试，并确保在维护、检
查和测试程序之后设备的持续安全。
如果适用，制造商的文件应说明不能用额定值不适当的线缆来替换可拆卸的电网电源线。
对于使用可更换电池的设备，应说明该特定电池的型号。
制造商应规定只能由制造商或其代理机构才能检查或提供的任何零部件。
对可更换的熔断器的额定值和特性应作出说明。
如果设备能够维修，必须保证安全维修和维修后设备仍然安全，因此应为维修人员提供如下内容的GB4793-××××
6说明：
a)可能影响维修人员的产品特定的风险；
b)对这些风险的防护措施；
c)设备修理后对其安全状态的验证。
5.4.6系统集成或由于特定条件的影响
由系统集成产生的情况或由特定环境或应用条件所导致的影响应在文件中加以描述。
6防电击
6.1概述
设备在正常条件和单一故障条件下均应保持防电击，可触及零部件不应出现危险带电。可触及零部
件和地之间或设备同一部分上距离在1.8m内（沿表面或通过空气）的任何两个可触及零部件之间的电
压、电流、电荷或能量不应超过6.3.1中正常条件下的限值，也不能超过6.3.2中单一故障条件下的限值。
电击防护措施可接受的方案见图1。应提供a)、b)、c)或d)中的一种作为初级防护措施的附加防护。
另一种方法是应采用e)或f)中一种单一的防护措施。
a)保护连接（见6.5.1）；
b)附加绝缘（见6.5.2）；
c)自动断开电源（见6.5.5）；
d)限流或限压装置（见6.5.6）；
e)加强绝缘（见6.5.3）；
f)保护阻抗（见6.5.4）。
图1电击防护措施可接受的方案双
重
绝
缘
6.5.6
限流或限压装
置6.4.2)
基本绝缘6.4.3
阻抗
6.5.5
自动断开电源6.5.1
保护连接6.5.2
附加绝缘危险零部件
可触及零部件6.5.4
保护阻抗6.5.3
加强绝缘6.4.1
外壳、保护挡板
和GB4793-××××
76.2例外
如果因操作原因，对下列零部件不可能做到既要防止可触及又要防止危险带电，则允许这些零部件
在危险带电时，操作人员在正常使用中是可触及的：
a)灯泡拆开之后的灯泡零部件和灯座；
b)预定要由操作人员更换的零部件（如电池），它们在更换时或在操作人员的其他操作行为时可
能是危险带电的，但只有在仅用工具才能可触及，而且标有警告标志。
如果a)和b）中的任何零部件从内部电容器接受电荷，则在断开电源10s后，这些零部件不应危
险带电。
6.3危险带电的限值
6.3.1正常条件下的限值
如果测量电压值超过a)的限值,同时其他测量值大于下列b)或c)中的限值，则被认为是危险带电：
a)交流电压限值为有效值30V和峰值42.4V，直流电压限值为60V。对预定在潮湿场所使用的
设备，交流电压限值为有效值16V和峰值22.6V，直流电压限值为35V；
b)电流限值为：
1)对正弦波电流为有效值0.5mA，对非正弦波或混合频率电流为峰值0.7mA，或者直流值2
mA。
2)涉及较高频率下可能的灼伤时，使用特殊测量电路，有效值70mA。
c)电容的电荷或能量限值为：
1)对电压不大于峰值15kV或直流15kV，电荷45μC；
2)对电压大于峰值15kV或直流15kV，贮存能量350mJ。
6.3.2单一故障条件下的限值
如果测量电压值超过a)的限值,同时其他测量值大于下列b)或c)中的限值，则被认为是危险带电：
a)交流电压限值为有效值50V和峰值70V，直流电压限值为120V。对预定在潮湿场所使用的
设备，交流电压限值为有效值33V和峰值46.7V，直流电压限值为70V。
b)电流限值为：
1)对正弦波电流为有效值3.5mA，对非正弦波或混合频率电流为峰值5mA，或者直流值15mA；
2)涉及较高频率下可能的灼伤时，使用特殊测量电路，有效值500mA。
c)电容限值见图2中线B。GB4793-××××
8
说明：
A——正常条件；
B——单一故障条件。
图2正常条件和单一故障条件下电容限值对应电压
6.4基本防护措施
6.4.1外壳和保护挡板
如果外壳或保护挡板用绝缘来提供防护，则它们应满足基本绝缘的要求。
如果外壳或保护挡板用限制接触来提供防护，则可触及零部件和危险带电零部件之间的电气间隙和
爬电距离应满足6.6的要求和基本绝缘适用的要求。
6.4.2基本绝缘
形成基本绝缘的电气间隙、爬电距离以及固体绝缘应满足6.6的要求。
6.4.3阻抗
用于初级防护措施的阻抗应满足下列所有要求：
a)应限制电流或电压至不超过6.3.2中适用的限值；
b)额定值应符合最大工作电压和消耗功率；
c)阻抗端子之间的电气间隙和爬电距离应满足6.6中基本绝缘适用的要求。
6.5附加防护措施
6.5.1保护连接
6.5.1.1概述
如果可触及导电零部件在6.4规定的初级防护措施出现单一故障时可能变为危险带电，则可触及导
电零部件应连接到保护导体端子，另一种方法是应采用与保护导体端子相连的导电保护屏蔽层将这些可
触及零部件与危险带电零部件隔离。GB4793-××××
96.5.1.2保护连接的完整性
应按照如下规定来保证保护连接的完整性：
a)保护连接应由直接的结构件，或独立的导体或者这二者组成；
b)对承受机械应力的焊接连接应采用与焊接无关的方法进行机械固定。这种连接不应用于其他目
的，例如固定结构件；
c)螺丝连接件应紧固防止松动；
d)如果设备的某一部分可由操作人员来拆除，则不能使设备剩余部分的保护连接断开；
e)可移动的导电的连接件，不应成为唯一的保护连接通道，除非将它们专门设计成供电气互连用；
f)电缆的外部金属编织物即使与保护导体端子连接也不应认为是保护连接；
g)如果由电网电源供电的电源通过设备供其他设备使用，则还应采取措施，使保护导体通过该设
备来保护其他设备。通过该设备的保护导体通路的阻抗不应超过6.5.1.4的规定值；
h)保护导体可以是裸导体也可以是绝缘导体，绝缘的颜色应是黄绿色，但下列情况除外：
1)对接地编织线，可以是黄绿色的也可以是无色透明的；
2)对内部保护导体以及和组件中的保护导体端子连接的其他导体，如果不可能因保护导体无
标识而引起危险，则可以使用任何颜色。
使用保护连接的设备应提供满足6.5.1.3要求的端子并应能适用于保护导体的连接。
6.5.1.3保护导体端子
保护导体端子应满足如下要求：
a)接触表面应为金属表面；
b)器具输入插座的整体式保护导体连接端应认为是保护导体端子；
c)对装有可拆卸软线的设备以及对永久性连接式设备，其保护导体端子应位于电网电源端子的近
旁；
d)如果设备不需要与电网电源相连，但仍然具有需要保护接地的电路或零部件，则保护导体端子
应位于需保护接地的该电路端子的附近。如果该电路有外部端子，则保护导体端子也应位于外
部；
e)电网电源电路的保护导体端子其载流能力至少应与电网电源供电端子的载流能力相当；
f)组合有其他端子的以及预定不使用工具连接和断开的插入式保护导体端子应设计成使保护导
体连接相对于其他连接最先接通和最后断开；
g)如果保护导体端子还要用于其他连接目的，则应首先用于连接保护导体，而且固定保护导体应
与其他连接无关。保护导体的连接方式应确保不可能由于进行不涉及保护导体的维修而将保护
导体拆除；
h)对需要用保护导体来对测量电路的单一故障提供防护的设备，应采用下列要求；
1)保护导体端子和保护导体至少应具有测量端子的电流额定值；
2)所装有的任何开关或断路装置不应断开保护连接；
i)功能接地端子（例如测量接地端子）应提供独立于保护导体连接的连接；
j)如果保护接地端子是一种连接螺钉组件(见图3)，则该螺钉应适合连接导体的尺寸，但其螺纹
尺寸不小于4mm，并至少应能啮合3圈螺纹；
k)保护连接所需的接触压力应不会由于构成连接部分的材料变形而减小。
说明：GB4793-××××
10A——固定的零部件；B——垫圈或夹板；C——防伸展装置；D——导体空间。
图3连接螺钉组件示例
6.5.1.4插头连接设备的保护连接阻抗
保护导体端子与规定要采用保护连接的每一个可触及零部件之间的阻抗不应超过0.1。如
果设备有不可拆卸的电源线，则电源线保护导体插头引脚与保护连接的每一个可触及零部件之
间的阻抗不应超过0.2。
6.5.1.5永久性连接式设备的保护连接阻抗
永久性连接式设备的保护连接应是低阻抗连接。
如果设备在电源的所有极上装有过流保护装置，以及如果在单一故障条件下过流保护装置电源一侧
的导线不可能变成与可触及导电零部件相连，则试验电流不必大于内部过流保护装置额定电流的2倍。
6.5.1.6变压器保护连接屏蔽层
如果变压器配有保护连接作用的屏蔽层，并且屏蔽层与连接到危险带电电路的绕组之间仅通过基本
绝缘分隔开，则屏蔽层应满足6.5.1.2a)和b)的要求，而且具有低阻抗。
6.5.2附加绝缘
电气间隙、爬电距离以及形成附加绝缘的固体绝缘应满足6.6中适用的要求。
6.5.3加强绝缘
电气间隙、爬电距离和组成加强绝缘的固体绝缘应满足6.6中适用的要求。
6.5.4保护阻抗
保护阻抗应在正常条件下将电流和电压限制在6.3.1的限值内，在单一故障条件下将电流和
电压限制在6.3.2的限值内。
保护阻抗端子之间的绝缘应满足6.7中双重绝缘或加强绝缘的要求。
保护阻抗应是下列规定的一种或一种以上的类型：
a)合适的单一元器件，其结构、选取和测试应能保证防电击的安全性和可靠性。特别地，元器件
应：
1)额定值为最大工作电压的2倍；
2)如果是电阻，则额定功率为最大工作电压下消耗功率的2倍。
b)元器件的组合。
保护阻抗不应是在真空、气体或半导体内使用电子导电的单一电子装置。
6.5.5自动断开电源
自动断开装置应满足下列两个要求：
a)自动断开装置的额定特性应规定成能在图2规定的时间范围内断开负载；
b)自动断开装置的额定值应与设备的最大额定负载条件相适应。
6.5.6限流或限压装置
限流或限压装置应满足下列所有要求：
a)限流或限压装置的额定特性应能限制电流或电压不超过6.3.2的限值；
b)限流或限压装置的额定值应与最大工作电压和（如果适用）最大工作电流相适应；
c)限流或限压装置端子间的电气间隙和爬电距离应满足6.6中附加绝缘适用的要求。
6.6绝缘要求
6.6.1绝缘的性质
6.6.1.1概述
电路和可触及零部件间或者单独的电路之间的绝缘由电气间隙、爬电距离和固体绝缘的组
合构成。当被用来提供防危险的保护时，绝缘必须能承受由出现在电网电源或者设备里可能出
现的电压造成的电应力。
6.6.1.2电气间隙GB4793-××××
11如果设备被规定能在超过2000m的海拔工作，则其电气间隙要乘以表2适用的系数。
表2海拔5000m内的电气间隙倍增系数
额定工作海拔高度
m倍增系数
≤2000
2001~3000
3001~4000
4001~50001.00
1.14
1.29
1.48
6.6.1.3爬电距离
根据CTI值，材料被分成如下的四组：
材料组别I600≤CTI
材料组别Ⅱ400≤CTI＜600
材料组别Ⅲa175≤CTI＜400
材料组别Ⅲb100≤CTI＜175
对于不知道CTI值的材料，假定为Ⅲb。
对玻璃、陶瓷或其他不产生漏电起痕的无机绝缘材料,没有爬电距离的要求。
6.6.1.4固体绝缘
固体绝缘是不可恢复的介质，损坏会在设备的生命周期里累积。固体绝缘同样会在重复的高压测试
中老化和降级。
6.6.1.5基于电路类型的绝缘要求
特殊类型电路的绝缘要求规定如下：
a)标称电源电压不超过300V的过电压类别Ⅱ的电网电源电路见6.6.2。
b)仅通过变压器的方式与a)中电路隔离的二次电路见6.6.3。
c)过电压类别Ⅲ或Ⅳ或者超过300V的过电压类别Ⅱ的电网电源电路见第A.1章。
d)仅通过变压器的方式与c)中电路隔离的二次电路见第A.2章。
e)具有一个或者多个下列特征的电路见第A.3章：
1)最大可能瞬态过电压受供电电源限制或者在设备内限制到低于电网电源电路的规定值的
已知值；
2)最大可能瞬态过电压超过电网电源电路规定值；
3)工作电压是超过一个电路的电压的总和，或者是一个混合电压；
4)工作电压包括一个重复峰值电压，该重复峰值电压包括一个周期性非正弦波形或者规律性
发生的非周期波形；
5)工作电压的频率超过30kHz。
6.6.2标称电源电压不高于300V的过电压类别Ⅱ的电网电源电路的绝缘
6.6.2.1电气间隙和爬电距离
电网电源电路的电气间隙和爬电距离应满足表3中的值，同时应考虑：
a)表3是基本绝缘和附加绝缘的值，对于加强绝缘，其值应是基本绝缘的2倍。
b)对于基本绝缘、附加绝缘和加强绝缘，污染等级3时的最小电气间隙是0.8mm。
c)如果设备的额定工作海拔高度高于2000m，电气间隙应乘以表2的倍增系数。
表3不超过300V的过电压类别Ⅱ的电网电源电路的电气间隙和爬电距离GB4793-××××
12相线-中线
电压
交流有效值
或直流值电气间
隙值爬电距离值
印制线路板材料 其他绝缘材料
污染等
级1污染等
级2污染等
级1污染等级2 污染等级3
所有材
料组别材料组
别Ⅰ，
Ⅱ，Ⅲa所有材
料组别材料组
别Ⅰ材料组
别Ⅱ材料组
别Ⅲa材料组
别Ⅰ材料组
别Ⅱ材料组
别Ⅲ
V mm mm mm mm mm mm mm mm mm mm
≤150 0.50.50.50.50.81.11.62.02.22.5
＞150～≤
3001.51.51.51.51.52.13.03.84.14.7
允许对爬电距离进行线性插值法。
6.6.2.2固体绝缘
6.6.2.2.1概述
在设备的预期寿命期间，在额定的环境条件下（见第4章），电网电源电路的固体绝缘应能承受可
能在正常使用中出现的电应力和机械应力。
固体绝缘还应满足如下的要求，如适用：
a)对于模制和模压零部件，6.6.2.2.2的要求；
b)印制线路板的内层，6.6.2.2.3的要求；
c)对于薄膜绝缘，6.6.2.2.4的要求。
6.6.2.2.2模制和模压零部件
对于基本绝缘、附加绝缘和加强绝缘，位于模压在一起的相同的两个层的界面上的导体，在模制完
成后应至少隔开0.4mm（见图4，L）。
说明：
1——层1；
2——层2；
C——导体；
L——导体间距。
图4两层间界面上的导体的间距
6.6.2.2.3印制线路板的内部绝缘层
对于基本绝缘、附加绝缘和加强绝缘，位于相同的两个层上的导体至少隔离0.4mm（见图5，L）。
GB4793-××××
13说明：
L——相邻导体的间距；
A——层；
C——导体。
图5沿着两个内层界面的相邻导体的间距
印制线路板的内部绝缘层的加强绝缘也应使各层有足够的介电强度。应使用下面的方法之一：
a)绝缘的厚度至少是0.4mm。
b)绝缘至少由印制线路板材料的两个独立的层组成，任一层的介电强度值都会被材料制造商规定
成不小于表4中基本绝缘的试验电压值。
c)绝缘至少由印制线路板材料的两个独立的层组成，并且各层组合的介电强度值都会被材料制造
商规定成不小于表4中加强绝缘的试验电压值。
表4不超过300V的过电压类别Ⅱ的电网电源电路的固体绝缘的试验电压
相线-中线电压
交流有效值或直流值1min交流试验电压 1min直流试验电压
基本绝缘和附加绝缘 加强绝缘 基本绝缘和附加绝缘 加强绝缘
V V V V V
≤150 1350 2700 1900 3800
＞150～≤300 1500 3000 2100 4200
6.6.2.2.4薄膜绝缘
对于基本绝缘、附加绝缘和加强绝缘，位于两个相同层（见图6，L）之间的导体应被隔离开，间
距应至少是6.6.2.1中的适用的电气间隙和爬电距离。
说明：
L——相邻导体的间距；
A——薄膜材料层，比如带子和聚酯薄膜；
C——导体。
注：层与层之间可能有空气存在。
图6相同层之间的相邻导体的间距
通过薄膜绝缘层的加强绝缘应有足够的介电强度。可以使用下述方法之一：
a)绝缘的厚度至少是0.4mm。
b)绝缘至少由薄膜材料的两个独立的层组成，任一层的介电强度值都会被材料制造商规定成不小
于表4中基本绝缘的试验电压值。
c)绝缘至少由薄膜材料的三个独立的层组成，任意两层已经过试验，具有足够的电气强度。
6.6.3由不超过300V的过电压类别Ⅱ的电网电源电路供电的二次电路的绝缘
6.6.3.1概述GB4793-××××
14本文件中，二次电路和电网电源电路的隔离是通过变压器实现的。在变压器内部，初级线圈和次级
线圈被加强绝缘、双重绝缘或者连接到保护导体端子的屏蔽层隔离。
6.6.3.2电气间隙
二次电路的电气间隙应满足a)或b)：
a)对于基本绝缘和附加绝缘，满足表5的值，或者对于加强绝缘，满足表5值的2倍；
b)使用表5中适用的试验电压，通过电压试验。
如下调整适用：
1)加强绝缘的试验电压值是基本绝缘的1.6倍；
2)如果设备的额定工作海拔高度高于2000m，电气间隙的值应乘以表2中适用的系数；
3)污染等级2的最小电气间隙是0.2mm，污染等级3的最小电气间隙是0.8mm。GB4793-××××
15表5不超过300V的过电压类别Ⅱ的由电网电源电路供电的二次电路的电气间隙和试验电压
二次电路工作电压 电网电压，相线-中线，过电压类别Ⅱ
≤150V
交流有效值＞150V～≤300V
交流有效值
交流有效值
V直流或交流峰值
V电气间隙
mm试验电压(交流有
效值)
V电气间隙
mm试验电压(交流有
效值)
V
16
30
50
100
150
300
600
1000
1250
1600
2000
2500
3200
4000
5000
6300
8000
10000
12500
16000
20000
25000
32000
40000
50000
6300022.6
42.4
70
140
210
420
840
1400
1750
2240
2800
3500
4480
5600
7000
8820
11200
14000
17500
22400
28000
35000
44800
56000
70000
882000.10
0.11
0.12
0.13
0.16
0.39
1.01
1.92
2.50
3.39
4.49
6.02
8.37
10.9
14.0
18.2
23.9
30.7
39.6
52.5
67.9
87.9
117
151
196
258500
510
520
540
580
770
1070
1630
1960
2390
2890
3520
4390
5320
6590
8270
10400
12900
16100
20400
25300
31600
40400
50300
62800
794000.48
0.50
0.53
0.61
0.69
0.94
1.61
2.52
3.16
4.11
5.30
6.91
9.16
11.6
14.9
19.1
24.7
31.6
40.5
53.5
68.9
89.0
118
153
198
260830
840
860
900
940
1040
1450
1970
2280
2730
3230
3850
4660
5610
6960
8620
10700
13300
16400
20700
25600
32000
40700
50800
63400
80000
允许线性内插值法。
6.6.3.3爬电距离
二次电路的基本绝缘或附加绝缘的爬电距离，应依据加在绝缘上的工作电压来满足表6中相应的值。
加强绝缘的值为基本绝缘的2倍。
表6二次电路的爬电距离GB4793-××××
16二次电路
工作电压
交流有效
值或直流印制线路板的材料 其他绝缘材料
污染等级
1污染等级
2污染等级1 污染等级2 污染等级3
所有材料
组材料组别
I,Ⅱ,或
Ⅲa所有材料
组材料组别
I材料组别
Ⅱ材料组别
Ⅲ材料组别
I材料组别
Ⅱ材料组别
Ⅲb
V mm mm mm mm mm mm mm mm mm
10
12.5
16
20
25
32
40
50
63
80
100
125
160
200
250
320
400
500
630
800
1000
1250
1600
2000
2500
3200
4000
5000
6300
8000
10000
12500
16000
20000
25000
32000
40000
50000
630000.025
0.025
0.025
0.025
0.025
0.025
0.025
0.025
0.040
0.063
0.10
0.16
0.25
0.40
0.56
0.75
1.0
1.3
1.8
2.4
3.2a0.04
0.04
0.04
0.04
0.04
0.04
0.04
0.04
0.063
0.10
0.16
0.25
0.40
0.63
1.0
1.6
2.0
2.5
3.2
4.0
5.0a0.08
0.09
0.10
0.11
0.125
0.14
0.16
0.18
0.20
0.22
0.25
0.28
0.32
0.42
0.56
0.75
1.0
1.3
1.8
2.4
3.2
4.2
5.6
7.5
10.0
12.5
16
20
25
32
40
50
63
80
100
125
160
200
2500.40
0.42
0.45
0.48
0.50
0.53
0.56
0.60
0.63
0.67
0.71
0.75
0.80
1.00
1.25
1.60
2.0
2.5
3.2
4.0
5.0
6.3
8.0
10.0
12.5
16
20
25
32
40
50
63
80
100
125
160
200
250
3200.40
0.42
0.45
0.48
0.50
0.53
0.80
0.85
0.90
0.95
1.00
1.05
1.1
1.4
1.8
2.2
2.8
3.6
4.5
5.6
7.1
9,0
11
14
18
22
28
36
45
56
71
90
110
140
180
220
280
360
4500.40
0.42
0.45
0.48
0.50
0.53
1.10
1.20
1.25
1.3
1.4
1.5
1.6
2.0
2.5
3.2
4.0
5.0
6.3
8.0
10.0
12.5
16
20
25
32
40
50
63
80
100
125
160
200
250
320
400
500
6001.00
1.05
1.10
1.20
1.25
1.3
1.4
1.5
1.6
1.7
1.8
1.9
2.0
2.5
3.2
4.0
5.0
6.3
8.0
10.0
12.5
16
20
25
32
40
50
63
80
100
1251.00
1.05
1.10
1.20
1.25
1.3
1.6
1.7
1.8
1.9
2.0
2.1
2.2
2.8
3.6
4.5
5.6
7.1
9.0
11
14
18
22
28
36
45
56
71
90
110
1401.00
1.05
1.10
1.20
1.25
1.3
1.8
1.9
2.0
2.1
2.2
2.4
2.5
3.2
4.0
5.0
6.3
8.0
10.0
12.5
16
20
25
32
40
50
63
80
100
125
160
允许线性内插值法。
a)电压超过1000V,对同样材料组别,在印制线路板材料上的爬电距离和在其他绝缘材料上的爬电距离是相同的。
b)对于电压超过630V,污染等级3,不推荐使用材料组别Ⅲb。
6.6.3.4固体绝缘
6.6.3.4.1概述
在设备预期寿命期间,在所有额定环境条件（见第4章）下，二次电路的固体绝缘应能承受可能发
生在正常使用中出现的电应力和机械应力。
如果适用，固体绝缘还需满足以下要求:
1)对模制和模压零部件,按6.6.3.4.2的要求；GB4793-××××
172)对印制线路板的内绝缘层,按6.6.3.4.3的要求；
3)对薄膜绝缘，按6.6.3.4.4的要求。
6.6.3.4.2模制和模压零部件
对基本绝缘、附加绝缘和加强绝缘,位于两个相同层之间的导体隔离(见图4，L)不小于表7中适
用的最小距离。
表7距离或厚度最小值(见6.6.3.4.2～6.6.3.4.4)
交流峰值或直流
工作电压或重复峰值电压最小值 交流峰值电压或直流
工作电压或重复峰值电压最小值
kV mm kV mm
＞0.0467～≤0.33 0.05 ＞8.0～≤10 3.5
＞0.33～≤0.8 0.1 ＞10～≤12 4.5
＞0.8～≤1.0 0.15 ＞12～≤15 5.5
＞1.0～≤1.2 0.2 ＞15～≤20 8
＞1.2～≤1.5 0.3 ＞20～≤25 10
＞1.5～≤2.0 0.45 ＞25～≤30 12.5
＞2.0～≤2.5 0.6 ＞30～≤40 17
＞2.5～≤3.0 0.8 ＞40～≤50 22
＞3.0～≤4.0 1.2 ＞50～≤60 27
＞4.0～≤5.0 1.5 ＞60～≤80 35
＞5.0～≤6.0 2 ＞80～≤100 45
＞6.0～≤8.0 3
6.6.3.4.3印制线路板的内绝缘层
对基本绝缘、附加绝缘和加强绝缘,位于两个相同层之间的导体隔离(见图5，L)不小于表7中适
用的最小距离。
印制线路板内绝缘层的加强绝缘,对各个单独的层也应有足够的介电强度。应使用下列方法之一:
a)绝缘厚度至少是表7规定的最小距离。
b)绝缘至少由印制线路板材料的两个独立的层组成，任一层的介电强度值都会被材料制造商规定
为不小于表5中基本绝缘的试验电压值。
c)绝缘至少由印制线路板材料的两个独立的层组成,由材料制造商规定的组合层的介电强度至少
满足表5试验电压的1.6倍。
6.6.3.4.4薄膜绝缘
对基本绝缘、附加绝缘和加强绝缘,位于两个相同层之间的导体隔离(见图6，L)不小于6.6.3.2
和6.6.3.3中适用的电气间隙和爬电距离。
使用薄膜绝缘层的加强绝缘应也有足够的介电强度。应使用下列方法之一:
a)绝缘厚度达到表7适用的最小值。
b)至少由两层分离的薄膜材料组成的绝缘,每层都由材料制造商规定为不小于表5中基本绝缘的
试验电压值。
c)至少由三层分离的薄膜材料组成的绝缘,任何两层之间都需要试验且要有足够的介电强度。
7防机械危险
7.1概述
在正常条件下或不易被察觉的单一故障条件下操作不应导致机械危险。机械危险包括但不仅限于：
a)可能导致割伤的锋利边缘（见7.2）；GB4793-××××
18b)可能挤压身体部位或刺穿皮肤的运动零部件（见7.3）；
c)可能在使用中或移动中跌落到人身上的不稳定的设备（见7.4）；
d)由于搬运装置（见7.5）、墙壁安装支架（见7.6）或其他支撑零部件（见7.5）毁坏而导致
的设备跌落；
e)从设备飞散的零部件（见7.7）。
7.2锋利边缘
设备所有易接触的零部件应光滑无棱角，避免在正常使用设备时造成伤害。
除非故障表现出明显的危险，设备上易接触的零部件在单一故障条件下不应造成任何伤害。
7.3运动零部件
7.3.1概述
运动零部件导致的危险不应超过除6.3.2规定外的可容许水平。7.3.4和7.3.5中规定的条件被认为
是可容许的限值。
7.3.2例外
如果由于操作原因不可能防止某些运动零部件引起潜在危险，则在下述情况下接触是允许的：
a)明显预定用于对设备外部零部件或材料进行加工的易接触的运动零部件的设备，例如：钻孔设
备和搅拌设备。这类设备应设计成能使无意中接触这种运动零部件的可能性减小到最低的限度
（如安装防护装置或提手等）。
b)除正常使用外，在进行常规维护时，如果由于技术上无法避免的原因，操作人员需接触可能会
引起危险的运动零部件才能完成某种操作，则如果采取了下列的所有措施，接触是允许的：
1)不用工具就不可能接触运动零部件；
2)给责任者的说明要包括一项声明，即操作人员必须经过培训才能允许进行带有危险性的操
作；
3)在接触运动零部件之前必须先行拆除的任何盖子或零部件上要有警告标志，标明操作人员
未经培训，禁止接触设备。作为替代方法，在盖子或零部件上标志表1序号13的符号，
并且在文件中给出警告说明。
7.3.3对身体部位构成机械危险的风险评定
综合考虑危险严重程度、暴露可能性和避免危险的可能性，应按照表2中描述最低防护措施将风险
降低到可容许水平。GB4793-××××
19表2对身体部位构成机械危险的防护措施
机械危险条件 最低防护措施d
严重程度a暴露的可能性b避免危险的可能性c
S E2 P2 C
S E2 P1 C
S E1 P2 C
S E1 P1 B
M E2 P2 B
M E2 P1 A
M E1 P2 A
M E1 P1 无
a严重程度：
M——中度危害，足以碰伤或划伤身体某部位。
S——严重危害，足以使身体某部位骨折或伤残。
b暴露的可能性：
E1——在正常使用过程中无意暴露。
E2——在正常使用过程中有意暴露。
c避免危险的可能性：
P1——可能避免
运动是可视的且速度足够低使身体部位远离且不被困住。
或在身体部位被困住之前启动可视或声音报警。
P2——不可避免
除P1之外的条件。
d最低防护措施：
A——低级措施：警示标志、可听或可视信号、使用说明。
B——中等措施：紧急开关、仅用工具才能移除的保护挡板或盖子、距离（见ISO13857）或隔离（见ISO13854
或EN349）。
C——严格措施：联锁、仅用工具才能移除的保护挡板或盖子，和从电源处切断的说明。
7.3.4压力和压强的限制
在正常条件和单一故障条件下应满足下面的限值。
最大允许的连续接触压强是50N/cm2，最大压力是150N。
至少3cm2的身体接触面积上的最大允许的短时压力是250N，最大持续时间是0.75s。
7.3.5运动零部件之间的间隙限制
7.3.5.1运动零部件之间的间隙限制－正常允许进入
如果身体部位能够插入到运动零部件之间，且运动零部件的力和压力超出了7.3.4的限值，在正常
条件和单一故障条件下，间隙的宽度不应从大于表3对于身体部位的最小间隙值减小到小于最小间隙值。
7.3.5.2运动零部件之间的间隙限制－正常防止进入
在正常条件和单一故障条件下，当零部件运动时，身体部位能插入进去的运动零部件之间的间隙不
应增加到超过表4的可接受的间隙。
表3防止不同身体部位挤压的最小保持间隙GB4793-××××
20身体部位 避免挤压的最小间隙a/mm 图例
躯干 500a
头 300a
腿 180
a
脚 120
a
脚趾 50
a50 max.
手臂 120a
手、腕、拳 100a
手指 25a
本表中的值是对于成年人的。对于儿童或者青少年操作的设备，应考虑更小的间隙。
表4防止不同身体部位进入的最大间隙
身体部位 防止接近的最大间隙aa/mm
头 120
脚 35
手指 4GB4793-××××
21本表中的值是对于成年人的。对于可能被儿童或者
青少年操作的设备，宜考虑更小的间隙。
a间隙a的示例见表3。
7.4稳定性
在操作前不固定在建筑物结构上的设备和设备的组件，在正常使用时物理上应是稳定的。
如果提供措施来确保操作人员在拉开抽屉等操作后使设备仍能保持稳定性，则这些措施应是自动的
或者应标有利用措施的警告标志。
7.5提起和搬运用装置
7.5.1概述
质量等于或大于18kg的设备或零部件应装有供提起和搬运用的装置，或在制造商文件中作出说明。
7.5.2提手和把手
如果供搬运用的提手或把手是装在设备上或随同设备一起提供的，则它们应能承受设备重量4倍的
力。
7.5.3提起装置和支撑零部件
提起装置的零部件和支撑重物的零部件应有额定最大承载，或试验通过4倍于最大静态承载。
7.6墙壁安装
对预定要安装在墙上或天花板上的设备，其支架应能承受设备重量4倍的力。
7.7飞散的零部件
如果一旦零部件失效飞散开来，设备应能控制或者限制可能会引起危险的零部件的能量。
对飞散的零部件所采用的防护装置应不借助工具不能拆除。
8耐机械应力
当设备承受在正常使用时可能遇到的机械应力时不应引起危险。正常要求的能量防护等级为5J，在
满足以下判据的时候，小于5J但不低于1J也是允许的。
a)经制造商风险评定后证明较低等级是合理但；
b)当设备按照预定的应用方式安装完成后，不会轻易被未经授权的人员或者无关人员触碰；
c)在正常使用状态下，设备仅在偶尔的操作，如调整、编程或维护时，才会被触及；
d)设备上标识了对应于GB/T20138的IK代码或者表1序号13的符号，并且在附带的文件中阐
明额定能量等级及其试验方法。对于最低额定环境温度小于2℃的非金属外壳，规定的限值应
适用于最低额定环境温度，如果冲击能量在GB/T20138的IK值之间，任何IK标识都应选择
离其最近的较低值。
9防止火焰蔓延
9.1概述
在正常条件下或单一故障条件下，火焰不应蔓延到设备的外面。图4是说明符合性检验方法的流程
图。GB4793-××××
22
图4防止火焰蔓延要求的流程图
9.2消除或减少设备的引燃源
如果满足下列所有要求，则认为引燃的可能性和着火发生率已被减小到可容许的水平。
a)1）或2)：
1)按9.4的规定，限制设备的电路或零部件可获得的电压、电流和功率。GB4793-××××
232)不同电位的零部件之间的绝缘满足基本绝缘的要求，或能证明桥接绝缘不会导致引燃。
b)将有关可燃液体的任何引燃危险减小到9.5规定的可容许水平。
c)对设计产生热量的电路，进行单一故障条件下的试验未出现引燃。
9.3一旦出现着火，将火焰控制在设备内
9.3.1概述
如果设备满足下列之一的结构要求，则认为火焰蔓延到设备外面的可能性已被减小到可容许的水平。
a)需由操作人员持续维持开关激活状态，以控制设备通电。
b)设备和设备的外壳符合9.3.2的结构要求，而且符合9.5适用的要求。
9.3.2结构要求
应符合下列结构要求：
a)连接器和安装元器件的绝缘材料应具有GB/T5169.16规定的V-2或更优的可燃性等级，对印
制线路板的要求见14.6。
b)绝缘的导线和电缆应能阻止火焰的蔓延。
c)外壳应符合下列要求：
1)对不符合9.4限能电路要求的电路，如图6所示，其成5˚夹角范围内外壳的底部和侧面
应符合下列要求之一：
i)没有开孔；
ⅱ)由金属材料制成，且开孔符合表5的规定；
ⅲ)由金属隔离网制成，其网眼中心距离不超过2mm×2mm，金属丝的直径至少为0.45
mm；
ⅳ)挡板开孔符合图5的规定。
2)外壳及任何挡板或挡火板应用金属（镁除外）材料制成，或者用可燃性等级为GB/T5169.16
规定的V-1或更优的非金属材料制成。
3)外壳以及任何挡板或挡火板应具有足够的刚性。
表5外壳底部允许的开孔
最小厚度
mm开孔的最大直径
mm开孔的最小中心距
mm
0.66 1.14 1.70(233个孔/645mm2)
0.66 1.19 2.36
0.76 1.15 1.70
0.76 1.19 2.36
0.81 1.91 3.18(72个孔/645mm2)
0.89 1.90 3.18
0.91 1.60 2.77
0.91 1.98 3.18
1.00 1.60 2.77
1.00 2.00 3.00GB4793-××××
24Y
X1
2
说明：
1——挡板（可以位于外壳底部的下面）；
2——外壳底部。
Y≥2X但不小于25mm。
图5挡板
说明：
A——被认为是危险着火源的设备的零部件或元器件。如果它是未另外防护的，则是指整个零部件或元器件；如果
它是被外罩局部防护的元器件，则是指未防护的部分。
B——A的轮廓线在水平面上的投影。
C——斜线，用来划出结构符合9.3.2c)1)和9.3.2c)2)规定的外壳底部和侧面的最小区域。该斜线围绕A
的周边的每一个点，以及相对于垂线呈5°夹角投射，其取向要确保能划出最大的面积。
D——结构符合9.3.2c)1)规定的底部的最小区域。
图6结构符合9.3.2c)1)规定的外壳底部的区域
9.4限能电路
限能电路是符合下列所有判据的电路：
a)出现在电路中的电压不大于交流有效值30V和峰值42.4V，或者直流60V。GB4793-××××
25b)用下列之一的方法来限制出现在电路中的电流:
1)由自身限制或用阻抗限制最大可获得电流，使其不会超过表6的适用值；
2)过流保护装置限制电流，使其不会超过表7的适用值；
3)用调节网络限制最大可获得电流，使其在正常条件下或在调节网络中出现的一个故障的情
况下不会超过表6的相关规定值。
c)至少采用基本绝缘与能量值超过上述判据a)或b)的其他电路隔离。
如果使用过流保护装置，则该过流保护装置应是某种熔断器或某种不可调的非自复位机电装置。
表6最大可获得电流值的限值
开路输出电压(U或Û)
V最大可获得电流
A
交流有效值 直流 峰值a交流有效值或直流
U≤2 U≤2 Û≤2.8 50
2＜U≤12.5 2＜U≤12.5 2.8＜Û≤17.6 100/U
12.5＜U≤18.7 12.5＜U≤18.7 17.6＜Û≤26.4 8
18.7＜U≤30 18.7＜U≤60 26.4＜Û≤42.4 150/U
a峰值(Û)是为方便使用而提供，适用于非正弦波形的交流电和纹波超过10%的直流电。由于电
流的有效值与发热相关，故应确定最大可获得电流的有效值。
表7过流保护装置的值
出现在电路中的电位(U或Û)
V过流保护装置在不大于
120s后断开的电流b、c
A
交流有效值 直流 峰值a交流有效值或直流
U≤2 U≤2 Û≤2.8 62.5
2＜U≤12.5 2＜U≤12.5 2.8＜Û≤17.6 125/U
12.5＜U≤18.7 12.5＜U≤18.7 17.6＜Û≤26.4 10
18.7＜U≤30 18.7＜U≤60 26.4＜Û≤42.4 200/U
a峰值(Û)是为方便使用而提供，适用于非正弦波形的交流电和纹波超过10%的直流电。由于电流的有
效值与发热相关，故应确定最大可获得电流的有效值。
b该评估值是基于所规定的保护装置的时间－电流分断特性，与额定分断电流是有区别的。（例如
ANSI/UL248-14的5A熔断器，规定为10A在120s或更短时间熔断，而IEC60127的T型4A熔断
器，规定为8.4A在120s或更短时间熔断。）
c熔断器的分断电流与温度有关，如果熔断器邻近的周围温度明显高于室温，则温度的影响就必须加以
考虑。
9.5对装有或使用可燃液体设备的要求
装在设备内的或规定与设备一起使用的可燃液体在正常使用条件下或单一故障条件下不应导致火
焰蔓延。
如果满足下列之一的要求，则认为由可燃液体导致的危险已减小到可容许的水平。GB4793-××××
26a)在正常条件或单一故障条件下，可燃液体表面的温度和与可燃液体表面接触的零部件的温度要
限制在不超过t-25℃的温度下，其中t为可燃液体的燃点。
注：燃点是指将某种液体加热（按规定的条件）到使其表面的蒸汽和（或）空气混合物在施加和撤离外部火焰时能
使火焰维持至少5s的温度。
b)要将液体的量限制在不可能导致火焰蔓延的程度。
c)如果液体能被引燃，则火焰要受到控制，以防止火焰蔓延到设备的外面。应提供详细的使用说
明，规定足以减小危险的程序。
9.6过流保护
9.6.1概述
预定要由电网电源供电的设备应用熔断器、断路器、热切断器、阻抗限制电路或类似装置来保护，
防止设备出现故障时从电网获得过大的能量。
过电流保护装置供电侧相反极性的电网连接的零部件至少需要具有基本绝缘。
过电流保护装置不应安装在保护导体上，熔断器或单级断路器不应装在多相设备的中线上。
9.6.2永久性连接式设备
永久性连接式设备中的过流保护装置是可选的，如果设备不带有过流保护装置，则设备安装说明书
中应规定在建筑物设施中需要的过流保护装置的特性。
9.6.3其他设备
如果采用过流保护装置，则应装在设备内部。
10设备的温度限值和耐热
10.1对防灼伤的表面温度限值
在40℃的环境温度下，易接触表面的温度在正常条件下不应超过表8规定的值，在单一故障条件下
不应超过105℃。
设备的最高额定环境温度高于40℃时，其易接触表面的温度在正常条件下允许超过表8规定的值，
在单一故障条件下允许超过105℃，但是超过部分不能大于最高额定环境温度与40℃的差值。
用挡板来防护的，防止受到意外接触的表面不认为是易接触表面，只要该挡板不用工具不能被拆除
即可。
表8正常条件下表面温度限值
零部件限值
℃
1外壳或挡板的外表面（意外接触）
a)无涂层或阳极化处理的金属 65
b)有涂层的金属（油漆，非金属） 80
c)塑料 85
d)玻璃和陶瓷 80
e)正常使用条件下不太可能接触的小区域(<2cm2) 100
2旋钮和手柄（正常使用时会接触）
a)金属 55
b)塑料 70
c)玻璃和陶瓷 65
d)在正常使用时仅被短时间(1s~4s)抓握的非金属部件 70
10.2绕组的温度GB4793-××××
27如果因温度过高可能会导致危险，则绕组绝缘材料的温度在正常条件下和单一故障条件下不应超过
表9的规定值。
表9绕组绝缘材料的最高温度
绝缘等级
(见GB/T11021)正常条件
℃单一故障条件
℃
A
B
E
F
H105
130
120
155
180150
175
165
190
210
10.3耐热
当设备在环境温度40℃或最高额定环境温度（如果温度更高）下工作时，其电气间隙和爬电距离应
符合6.6的要求。
非金属材料的外壳应能耐高温。绝缘材料应有足够的耐热能力。
11防止流体和固体异物危险
11.1概述
设备的设计应能保护操作人员和周围区域免受正常使用中遇到的流体和固体异物的危害。
注：可能会遇到的流体分为三类：
a)持续接触的流体，如盛于容器中的流体；
b)偶然接触的流体，例如清洗用流体；
c)无意中（非预期）接触的液体，制造商无法对此类情况采取防护措施。
考虑制造商规定的流体，包括清洗和去污染流体。不考虑其他流体。
11.2清洗
如果制造商规定了清洗或去污处理流程，则该流程不应导致直接的危险、电气危险或者因腐蚀原因
或其他原因导致使保证安全的结构件强度降低的危险。
11.3洒落
如果正常使用时液体可能会洒落到设备中，则设备在设计上应确保不会发生危险，例如由于绝缘或
危险带电的内部无绝缘的零部件受潮带来的危险，或是由于设备零部件接触到潜在侵蚀性物质（如腐蚀
性的、有毒性的或者可燃的液体）所引起的危险。
11.4溢出
在正常使用时，从能过量注入液体的设备内任何容器中溢出的液体不应导致危险，例如由于绝缘或
危险带电的内部无绝缘的零部件受潮带来的危险。
在容器注满液体后可能要移动的设备应防止液体从容器中荡出。
11.5电池电解液
电池的安装应确保使电池电解液的泄漏不会损害安全。
11.6具有额定防护等级的设备（IP代码）
如果制造商规定设备符合GB/T4208定义的一种防护等级，设备防护固体异物和水的侵入的等级应
与对应的防护等级相适应。
11.7流体压力和泄漏
11.7.1最大压力
在正常使用或单一故障条件下，设备的零部件能承受的最大压力不应超过该零部件的额定最大工作
压力。GB4793-××××
28最大压力被认为是下列的最大值：
a)对外部压力源规定的额定最大供应压力；
b)作为设备一部分提供的过压安全装置的压力设定值；
c)除非压力已被过压安全装置限制，作为设备一部分的压力发生装置所产生的最大压力。
11.7.2高压泄漏和破裂
在正常使用时同时具有下列两个特性的、装有流体的零部件不应由于破裂或泄漏而导致危险。
a)压力和容积的乘积大于200kPa·L；
b)压力大于50kPa。
11.7.3低压单元的泄漏
装有流体的零部件在压力低于11.7.2条件下发生泄漏不应导致危险。
11.7.4过压安全装置
过压安全装置在正常使用时不应动作，它应符合下列要求：
a)连接在靠近预定要保护的系统中装有液体的零部件的附近；
b)容易接近，以便进行检查、维护和修理；
c)在不使用工具的条件下就不能对其进行调节；
d)压力释放孔的位置和方向应确保释放的物质不朝向任何人员；
e)压力释放孔的位置和方向应确保过压安全装置的动作不会在可能导致危险的零部件上沉积释
放的物质；
f)具有足够的释放能力，以确保压力不会超过系统的额定最大工作压力。
在过压安全装置和预定要保护的零部件之间不应装有截止阀。
12防辐射（包括激光源）、声压和超声压
12.1概述
设备应提供防内部产生的紫外线、电离辐射和微波辐射、激光源、以及声压和超声压效应的保护。
12.2产生电离辐射的设备
12.2.1电离辐射
12.2.1.1概述
包含或产生电离辐射（来自放射源或X射线）的设备应满足以下要求：
a)如果是以释放辐射为目的的，则应满足12.2.1.2中的要求。或者，如果在GB/T19661的范
围中，则应根据GB/T19661来试验、分类和标志；
b)如果是使用或产生辐射，但仅释放杂散辐射，则应满足12.2.1.3的要求。
12.2.1.2预定释放辐射的设备
包含放射性物质或产生X射线的设备，以及以向设备外释放电离辐射为目的的设备，如果在距离外
表面50mm的任何易于到达的点的有效剂量率超过5μSv/h，设备应标志如下所有的标记：
a)表1序号16的符号；
b)对于含有一种或多种放射性物质的设备，放射性元素的缩写；
c)1m处的最大剂量率的值，或当剂量率的值在1μSv/h和5μSv/h之间以m为单位适当的距
离。
示例：“1m处2.5μSv/h”;“0.3m处3μSv/h”。
12.2.1.3预定不释放辐射的设备
非预定杂散辐射在距离设备外表面100mm的任何易于到达的点的有效辐射剂量率应不超过1μ
Sv/h。
12.2.2加速电子
设备的结构应保证用超过5kV的电压来加速电子的隔离室不用工具就不能打开。GB4793-××××
2912.3光辐射
带有发射紫外线、可见光或红外辐射的灯和灯系统的设备（包括发光二极管），不应允许可能造成
危险的辐射意外泄漏。
12.4微波辐射
在正常条件和单一故障条件下，在距离设备50mm的任意一点，其频率在1GHz和100GHz之间的杂
散微波辐射的功率密度不应超过10W/m2。本要求不适用于预定传送微波辐射的设备的零部件，例如位
于波导输出端口。
12.5声压和超声压
12.5.1声压级
如果设备产生的噪声达到可能导致危害的等级，则制造商应测量设备能产生的最大声压级（但报警
产生的和位于远程的零部件产生的声音不包括在内）。
安装说明书应规定，在设备安装完后位于使用的位置上，责任者如何能确保设备产生的声压级不会
达到可能导致危害的值。这些说明应明确能迅速得到和切实可行的防护材料，或可能采用的一些措施，
包括安装隔音板或消声罩。
12.5.2超声压
如果不预定释放超声的设备产生的超声达到可能会引起危险的等级，则制造商应测量设备能产生的
最大超声压级。
如果以预定释放超声的设备产生的超声达到可能会引起危险的等级，则制造商应测量设备能产生的
最大超声压级。
外部的有用波束，超声压在频率20kHz和100kHz的范围内，不应超过20μPa基准压力值以上110dB。
如果有用波束内部的超声压在频率20kHz和100kHz的范围内超过110dB，那么设备应使用表1序号13
的符号加以标识；并且其文件中应包含以下内容：
a)有用波束的尺寸；
b)有用波束超声压超过110dB的区域；
c)在波束区域内部的最大声压值。
12.6激光源
使用激光源的设备应满足GB7247.1的要求。
13对释放的气体和物质、爆炸和内爆的防护
13.1有毒和有害气体和物质
设备不得在正常条件和单一故障条件下释放有害量的危险物质。
如果释放出潜在危险物质，则操作人员不应直接暴露于一定量的可能造成伤害的物质中。
如果设备的正常运行需要排放有害物质，并且如果排放意图由责任者按照制造商的说明进行管理，
则这种排放不被视为释放有害物质。
13.2爆炸和内爆
13.2.1元器件
如果因过热或过载易于引起爆炸的元器件，未装有压力释放装置，则在设备中应装有保护操作人员
的防护装置。
压力释放装置的位置应确保在卸荷时不会给操作人员造成危险。其结构应确保任何压力释放装置不
会被阻塞。
13.2.2电池和电池的充电
电池不应由于过度充电、放电或由于电池安装时极性不正确而引起爆炸或出现着火危险。如果有必
要，设备中应提供防护，除非制造商的说明书规定，该设备只能使用具有内部保护的电池。
如果由于装上错误型号的电池（例如，如果规定要装具有内部保护的电池）可能会引起爆炸或着火GB4793-××××
30危险，则应在电池舱、安装支架上或在其近旁标上警告标志，而且还应在制造商说明书中给出警告语句。
可接受的标志是表1序号13的符号。
如果设备具有能对可充电电池充电的装置，且如果不可充电电池有可能被安装和连接在电池舱内，
则应在电池舱内或其近旁标上警告标志。该标志应给出警告，防止对不可充电电池充电，同时还应标出
能与充电电路一起使用的可充电电池的型号。可接受的标志是表1序号13的符号。
电池舱的设计应做到不可能因可燃性气体的积聚而引起爆炸和着火。
13.2.3阴极射线管的内爆
对最大屏面尺寸超过160mm的阴极射线管，其自身应能防内爆影响和防机械撞击，除非其外壳提供
足够的防护。
自身无防护的阴极射线管应装有不用工具就不能拆卸的有效防护屏。如果使用玻璃的隔离屏，则它
不应与阴极射线管的表面接触。
当阴极射线管正确安装时无需再作附加防护，则认为这种阴极射线管自身具有对内爆影响的防护能
力。
14元器件和组件
14.1概述
如果涉及安全，则元器件和组件，应按其规定的额定值使用，除非已作出特定的例外规定。
14.2电动机
14.2.1电动机温度
当电动机堵转或阻止启动时出现电击危险、温度危险或燃烧危险，则应采用符合14.3要求的过温保
护装置或热保护装置来对电动机进行保护。
14.2.2串激电动机
如果串激电动机转速过高会引起危险，则它应直接接到由该串激电动机驱动的装置上。
14.3过温保护装置
过温保护装置是在单一故障条件下动作的装置，应符合下列所有要求：
a)在结构上应做到能保证功能可靠；
b)额定值能满足切断使用它们的电路中最大的电压和电流；
c)在正常条件下不动作。
如果在温度控制系统（如温控器）失效的情况下，使用自行复位的过温保护装置来防止危险发生，
那么设备被保护的部分在再次运行之前应需干预。
14.4熔断器座
对装有预定要由操作人员来更换熔断器的熔断器座在更换熔断器时应不能触及到危险带电零部件。
14.5电网电源电压选择装置
电网电源电压选择装置在结构上应做到不会意外发生将一个电压或一种类型电源转换到另一个电
压或另一种类型电源。见5.1.3d）中的标志要求规定。
14.6印制线路板
印制线路板应采用可燃性等级为GB/T5169.16的V-1或更优的材料。
14.7用于限制瞬态过电压的电路
构成设备一部分的任何过压限制电路应具有足够的强度，以限制可能出现的瞬态过电压。
15利用联锁装置的保护
15.1概述
用来防止操作人员遭受危险的联锁装置应在危险消除之前防止操作人暴露在危险中，并应符合15.2
和15.3的要求。GB4793-××××
3115.2防止重新启动
对保护操作人员的联锁装置，在引起联锁装置起作用的动作返回或取消之前，应能防止由于操作人
员不使用工具重新启动而再次引起危险。
15.3可靠性
保护操作人员的联锁装置应保证在设备的预期寿命期间不可能出现单一故障，或者不会引起危险。GB4793-××××
32附录A
（规范性）
6.6中未涵盖的绝缘要求
A.1电网电源电路的绝缘
A.1.1概述
如果制造商把设备的额定过电压类别定为Ⅲ或Ⅳ，则应适用本附录的相关要求。
过电压类别Ⅱ适用于预定由建筑物布线供电的设备。它适用于插头连接式设备及永久性连接式设备。
6.6仅包含了对标称电源电压不超过300V时过电压类别Ⅱ的要求。对更高过电压类别和标称电源电压超
过300V时过电压类别Ⅱ的要求见本附录。
过电压类别Ⅲ用于预定构成建筑物布线装置部分的设备。此类设备包括插座插孔、熔断器座和某些
电网电源设施控制设备。如果要求更高程度的可靠性和可用性，制造商也可设计过电压类别Ⅲ的设备。
过电压类别IV用于安装在建筑物供电电源源头或其附近，在建筑物入口和主配电盘之间的设备。此
类设备可能包括电表和初级过电流保护装置。如果要求更高程度的可靠性和可用性，制造商也可设计过
电压类别IV的设备。
A.1.2电网电源电路电气间隙和爬电距离
电网电源电路的电气间隙和爬电距离应满足下述表格中的适用数值：
a)标称电源电压超过300V的过电压类别Ⅱ的电网电源电路见表A.2；
b)过电压类别Ⅲ的电网电源电路见表A.3；
c)过电压类别Ⅳ的电网电源电路见表A.4。
表A.2～表A.4中的数值针对基本绝缘和附加绝缘。加强绝缘的数值应是基本绝缘数值的2倍。
如果设备的额定工作海拔高度高于2000m，电气间隙应乘以表A.1中适用的倍增系数。
表A.1额定在不超过5000m海拔下工作设备的电气间隙倍增系数
额定工作海拔
m倍增系数
≤2000
2001～3000
3001～4000
4001～50001.00
1.14
1.29
1.48
表A.2超过300V过电压类别Ⅱ的电网电源电路的电气间隙和爬电距离
相线-中线电
压交流有效
值或直流电气间
隙爬电距离
印制线路板材料 其他绝缘材料
污染等级
1污染等级2污染等级
1污染等级2 污染等级3
所有材料
组别材料组别
I,Ⅱ,
Ⅲa所有材料
组别材料组别
I材料组别
Ⅱ材料组别
Ⅲ材料组别
I材料组别
Ⅱ材料组别
Ⅲ
V mm mm mm mm mm mm mm mm mm mmGB4793-××××
33＞300～≤
6003.0 3.0 3.0 3.0 3.0 4.3 6.0 7.5 8.3 9.4
＞600～≤
10005.5 5.5 5.5 5.5 5.5 7.2 10.0 12.5 14.0 16.0
表A.3过电压类别Ⅲ的电网电源电路的电气间隙和爬电距离
相线-中线电
压交流有效
值或直流电气间
隙爬电距离
印制线路板材料 其他绝缘材料
污染等级
1污染等级
2污染等级
1污染等级2 污染等级3
所有材料
组别材料组别
I,Ⅱ,
Ⅲa所有材料
组别材料组
别I材料组别
Ⅱ材料组别
Ⅲ材料组别
I材料组别
Ⅱ材料组别
Ⅲ
V mm mm mm mm mm mm mm mm mm mm
≤150 1.5 1.5 1.5 1.5 1.5 1.5 1.6 2.0 2.2 2.5
＞150～≤
3003.0 3.0 3.0 3.0 3.0 3.0 3.0 3.8 4.1 4.7
＞300～≤
6005.5 5.5 5.5 5.5 5.5 5.5 6.0 7.5 8.3 9.4
＞600～≤
10008.0 8.0 8.0 8.0 8.0 8.0 10.0 12.5 14.0 16GB4793-××××
34表A.4过电压类别Ⅳ的电网电源电路的电气间隙和爬电距离
相线-中线电
压交流有效
值或直流电气间隙爬电距离
印制线路板材料 其他绝缘材料
污染等级
1污染等级
2污染等级
1污染等级2 污染等级3
所有材料
组别材料组别
I,Ⅱ,
Ⅲa所有材料
组别材料组
别I材料组别
Ⅱ材料组别
Ⅲ材料组别
I材料组别
Ⅱ材料组别
Ⅲ
V mm mm mm mm mm mm mm mm mm mm
≤150 3.0 3.0 3.0 3.0 3.0 3.0 3.0 3.0 3.0 3.0
＞150～≤
3005.5 5.5 5.5 5.5 5.5 5.5 5.5 5.5 5.5 5.5
＞300～≤
6008.0 8.0 8.0 8.0 8.0 8.0 8.0 8.0 8.3 9.4
＞600～≤
100014.0 14.0 14.0 14.0 14.0 14.0 14.0 14.0 14.0 16.0
A.1.3电网电源电路的固体绝缘
A.1.3.1概述
在设备的预期寿命期间，在所有额定的环境条件（见第4章）下，电网电源电路的固体绝缘应能承
受在正常使用时可能出现的电应力和机械应力。
如适用，固体绝缘还应满足以下要求：
1)对模制和模压零部件，A.1.3.2的要求；
2)对印制线路板的内层，A.1.3.3的要求；
3)对薄膜绝缘，A.1.3.4的要求。
A.1.3.2模制和模压零部件
对于基本绝缘、附加绝缘和加强绝缘，完成模压后，位于模压在一起的两个相同层之间的导体(见
图A.1，L)应至少间隔表A.5中规定适用的最小距离。
说明：
1——层1；2——层2；
C——导体；L——导体间距。
图A.1两层间界面上的导体的间距
A.1.3.3印制线路板内部绝缘层GB4793-××××
35对于基本绝缘、附加绝缘和加强绝缘，位于相同的两个层上的导体(见图A.2，L)应至少间隔表A.5
中规定的适用的最小距离。
说明：
L——相邻导体的间距；
A——层；
C——导体。
图A.2沿着两个内层界面的相邻导体的间距
表A.5固体绝缘间距或厚度最小值
相线-中线电压 最小厚度a最小间距La,b(见图A.2)
Vr.m.s.或Vd.c. mm mm
≤300 0.4 0.4
＞300～≤600 0.6 0.6
＞600～≤1000 1.0 1.0
a这些数值独立于过电压类别。
b这些数值适用于基本绝缘、附加绝缘和加强绝缘。
印制线路板内部绝缘层各层的加强绝缘应具有足够的介电强度，应采用下列方法之一：
a)绝缘厚度至少是表A.5中的规定数值；
b)绝缘至少由印制线路板材料的两个分离的层组成，任一层的介电强度值都会被材料制造商规定
成不小于表A.6、表A.7和表A.8中基本绝缘的试验电压值。
c)绝缘至少由印制线路板材料的两个分离的层组成，并且各层组合的介电强度值都会被材料制造
商规定成不小于表A.6、表A.7和表A.8中加强绝缘的试验电压值。
表A.6超过300V的过电压类别Ⅱ的电网电源电路固体绝缘的试验电压
相线－中线电压交
流有效值或直流
V试验电压
5s交流试验
Va.c.r.m.s.脉冲试验
VpeaA
基本绝缘和附加
绝缘加强绝缘基本绝缘和附
加绝缘加强绝缘
＞300～≤600 2210 3510 4000 6400
＞600～≤1000 3310 5400 6000 9600
表A.7过电压类别Ⅲ的电网电源电路固体绝缘的试验电压GB4793-××××
36相线-中线电压交流
有效值或直流
V试验电压
5s交流试验
Vr.m.s.脉冲试验
Vpeak
基本绝缘和附加
绝缘加强绝缘基本绝缘和附
加绝缘加强绝缘
≤150 1390 2210 2500 4000
＞150～≤300 2210 3510 4000 6400
＞300～≤600 3310 5400 6000 9600
＞600～≤1000 4260 7400 8000 12800
表A.8过电压类别Ⅳ的电网电源电路固体绝缘的试验电压
相线-中线电压交流
有效值或直流
V试验电压
5s交流试验
Vr.m.s.脉冲试验
Vpeak
基本绝缘和附加
绝缘 加强绝缘基本绝缘和附加
绝缘 加强绝缘
≤150 2210 3510 4000 6400
＞150～≤300 3310 5400 6000 9600
＞300～≤600 4260 7400 8000 12800
＞600～≤1000 6600 11940 12000 19200
A.1.3.4薄膜绝缘
对于基本绝缘、附加绝缘和加强绝缘，位于两个相同层（见图A.3，L）之间的导体应被隔离开，间
距应至少是A.1.2中的适用的电气间隙和爬电距离。
说明：
L——相邻导体的间距；A——薄膜材料层，比如带子和聚酯薄膜；C——导体。
层与层之间可能有空气存在。
图A.3相同两层之间相邻导体的间距
通过薄膜绝缘层的加强绝缘应具有足够的介电强度。应使用下列方法之一：GB4793-××××
37a)绝缘厚度至少应为表A.5的数值。
b)绝缘至少由薄膜材料的两个分离的层组成，任一层的介电强度值都会被材料制造商规定成不小
于表A.6、表A.7和表A.8中基本绝缘的试验电压值。
c)绝缘至少由薄膜材料的三个独立层组成，任意两层都要通过测试来证明拥有合适的介电强度。
A.2二次电路中的绝缘
A.2.1概述
本文件中，二次电路和电网电源电路的隔离是通过变压器实现的。在变压器内部，初级线圈和次级
线圈被加强绝缘、双重绝缘或者连接到保护导体端子的屏蔽层隔离。
A.2.2电气间隙
二次电路的电气间隙应满足a)或b)：
a)对于基本绝缘或附加绝缘，满足表A.9、表A.10和表A.11的值，或对于加强绝缘，满足该数
值的2倍；
b)采用表A.9、表A.10和表A.11中的适用数值，通过电压试验。
对于表A.9～表A.11的应用，需满足如下要求：
1)加强绝缘的试验电压值是基本绝缘的1.6倍；
2)如果设备的额定工作海拔高度高于2000m，电气间隙数值应乘以表A.1中适用的系数；
3)污染等级2的最小电气间隙为0.2mm，污染等级3的最小电气间隙为0.8mm。
表A.9超过300V的过电压类别Ⅱ的电网电源电路供电的二次电路的电气间隙和试验电压GB4793-××××
38二次电路工作电压 电网电源电压
＞300～≤600V
r.m.s.电网电源电压
＞600～≤1000V
r.m.s.
交流有效值.直流或交流峰值 电气间隙 试验电压 电气间隙 试验电压
V V mm Vr.m.s. mm Vr.m.s.
16
33
50
100
150
300
600
1000
1250
1600
2000
2500
3200
4000
5000
6300
8000
10000
12500
16000
20000
25000
32000
40000
50000
6300022.6
46.7
70
140
210
420
840
1400
1750
2240
2800
3500
4480
5600
7000
8820
11200
14000
17500
22400
28000
35000
44800
56000
70000
882001.5
1.5
1.5
1.6
1.6
1.8
2.4
3.5
4.2
5.2
6.5
8.1
10
12
16
20
26
33
42
55
71
91
120
154
199
2611390
1390
1390
1450
1450
1540
1620
2450
2770
3190
3700
4300
4950
5780
7400
8980
11200
13800
16900
21200
26300
32600
41600
52200
66100
853002.9
3.0
3.0
3.1
3.2
3.4
3.9
5.0
5.8
6.9
8.2
9.8
12
15
18
22
28
35
44
57
73
93
122
157
202
2621590
2210
2210
2260
2300
2400
2630
3110
3430
3850
4330
4920
5780
7000
8200
9700
11900
14500
17600
21900
27000
33200
42200
53100
67000
85600
允许线性内插值法。
表A.10由过电压类别Ⅲ的电网电源电路产生的二次电路的电气间隙和试验电压
二次电路工作电
压电网电源电压
≤150V
交流有效值电网电源电压
＞150～≤300V
交流有效值电网电源电压
＞300～≤600V
交流有效值电网电源电压
＞600～≤1000V
交流有效值
交流有
效值
V直流或
交流峰
值
V电气间隙
mm试验电压
交流有效
值电气间隙
mm试验电压
交流有效
值电气间隙
mm试验电压
交流有效
值电气间隙
mm试验电
压
交流有
效值
16
33
50
100
15022.6
46.7
70
140
2100.48
0.50
0.53
0.61
0.691100
1100
1120
1170
12001.5
1.5
1.5
1.6
1.61800
1800
1800
1880
18802.9
3.0
3.0
3.1
3.22820
2900
2900
2960
30205.4
5.4
5.5
5.6
5.74240
4240
4300
4360
4420GB4793-××××
39300
600
1000
1250
1600
2000
2500
3200
4000
5000
6300
8000
10000
12500
16000
20000
25000
32000
40000
50000
63000420
840
1400
1750
2240
2800
3500
4480
5600
7000
8820
11200
14000
17500
22400
28000
35000
44800
56000
70000
882000.94
1.6
2.5
3.2
4.1
5.3
6.9
9.2
12
15
19
25
32
41
54
69
89
118
153
198
2601360
1880
2500
3020
3560
4180
5040
6080
7500
9100
11200
14100
17400
21500
27200
33500
41600
53000
66100
82400
1040001.8
2.4
3.5
4.2
5.2
6.5
8.1
10
12
16
20
26
33
42
55
71
91
120
154
199
2612040
2440
3200
3620
4120
4800
5560
6400
7500
9600
11600
14600
17800
21900
27600
34300
42400
53700
66500
82700
1044003.4
3.9
5.0
5.8
6.9
8.2
9.8
12
15
18
22
28
35
44
57
73
93
122
157
202
2623140
3440
4000
4480
5040
5620
6320
7500
9100
10600
12600
15500
18700
22800
28400
35200
43200
54500
67600
83800
1047006.0
6.6
7.4
8.1
9.3
11
12
15
17
20
25
31
38
47
60
76
96
125
160
205
2654600
4860
5240
5560
6120
7000
7500
9100
10100
11600
14100
16900
20000
24200
29700
36400
44400
55600
68700
84900
105700
允许线性内插值法。
表A.11过电压类别Ⅳ的电网电源电路供电的二次电路的电气间隙和试验电压
二次电路工作电压 电网电源电压
≤150V
交流有效值电网电源电压
＞150～≤300V
交流有效值电网电源电压
＞300～≤600V
交流有效值电网电源电压
＞600～≤1000V
交流有效值
交流有
效值
V直流或交
流峰值
V电气间隙
mm试验电压
交流有效
值电气间隙
mm试验电压
交流有效
值电气间隙
mm试验电压
交流有效
值电气间隙
mm试验电压
交流有效
值
16
33
50
100
150
300
600
1000
1250
1600
2000
2500
3200
4000
5000
630022.6
46.7
70
140
210
420
840
1400
1750
2240
2800
3500
4480
5600
7000
88201.5
1.5
1.5
1.6
1.6
1.8
2.4
3.5
4.2
5.2
6.5
8.1
10
12
16
201800
1800
1800
1880
1880
2040
2440
3200
3620
4120
4800
5560
6400
7500
9600
116002.9
3.0
3.0
3.1
3.2
3.4
3.9
5.0
5.8
6.9
8.2
9.8
12
15
18
222820
2900
2900
2960
3020
3140
3440
4000
4480
5040
5620
6320
7500
9100
10600
126005.4
5.4
5.5
5.6
5.7
6.0
6.6
7.4
8.1
9.3
11
12
15
17
20
254240
4240
4300
4360
4420
4600
4860
5240
5560
6120
7000
7500
9100
10100
11600
141008.3
8.3
8.4
8.5
8.6
8.9
9.6
10
11
12
13
15
17
19
23
275680
5680
5740
5800
5860
5960
6240
6400
7000
7500
8100
9100
10100
11200
13100
15100GB4793-××××
408000
10000
12500
16000
20000
25000
32000
40000
50000
6300011200
14000
17500
22400
28000
35000
44800
56000
70000
8820026
33
42
55
71
91
120
154
199
26114600
17800
21900
27600
34300
42400
53700
66500
82700
10440028
35
44
57
73
93
122
157
202
26215500
18700
22800
28400
35200
43200
54500
67600
83800
10470031
38
47
60
76
96
125
160
205
26516900
20000
24200
29700
36400
44400
55600
68700
84900
10570033
40
50
63
79
99
129
164
209
26817800
21000
25500
31000
37600
45400
57100
70100
86300
106800
允许线性内插值法。
A.2.3爬电距离
二次电路的基本绝缘或附加绝缘的爬电距离，应根据施加在绝缘部分的工作电压，满足表A.12中的
合适数值。加强绝缘的爬电距离是基本绝缘的2倍。
表A.12二次电路的爬电距离GB4793-××××
41二次工作电
压
交流有效值
或直流印制线路板的材料 其他绝缘材料
污染等级
1污染等级
2污染等级
1污染等级2 污染等级3
所有材料
组材料组别
I,Ⅱ,或Ⅲa所有材料
组材料组别
I材料组别
I材料组别
Ⅱ材料组别
Ⅲb材料分组
Ⅱ材料分组
Ⅲb
V mm mm mm mm mm mm mm mm mm
10
12.5
16
20
25
32
40
50
63
80
100
125
160
200
250
320
400
500
630
800
1000
1250
1600
2000
2500
3200
4000
5000
6300
8000
10000
12500
16000
20000
25000
32000
40000
50000
630000.025
0.025
0.025
0.025
0.025
0.025
0.025
0.025
0.040
0.063
0.10
0.16
0.25
0.40
0.56
0.75
1.0
1.3
1.8
2.4
3.2a0.04
0.04
0.04
0.04
0.04
0.04
0.04
0.04
0.063
0.10
0.16
0.25
0.40
0.63
1.0
1.6
2.0
2.5
3.2
4.0
5.0a0.08
0.09
0.10
0.11
0.125
0.14
0.16
0.18
0.20
0.22
0.25
0.28
0.32
0.42
0.56
0.75
1.0
1.3
1.8
2.4
3.2
4.2
5.6
7.5
10.0
12.5
16
20
25
32
40
50
63
80
100
125
160
200
2500.40
0.42
0.45
0.48
0.50
0.53
0.56
0.60
0.63
0.67
0.71
0.75
0.80
1.00
1.25
1.60
2.0
2.5
3.2
4.0
5.0
6.3
8.0
10.0
12.5
16
20
25
32
40
50
63
80
100
125
160
200
250
3200.40
0.42
0.45
0.48
0.50
0.53
0.80
0.85
0.90
0.95
1.00
1.5
1.1
1.4
1.8
2.2
2.8
3.6
4.5
5.6
7.1
9.0
11
14
18
22
28
36
45
56
71
90
110
140
180
220
280
360
4500.40
0.42
0.45
0.48
0.50
0.53
1.10
1.20
1.25
1.3
1.4
1.5
1.6
2.0
2.5
3.2
4.0
5.0
6.3
8.0
10.0
12.5
16
20
25
32
40
50
63
80
100
125
160
200
250
320
400
500
6001.00
1.05
1.10
1.20
1.25
1.3
1.4
1.5
1.6
1.7
1.8
1.9
2.0
2.5
3.2
4.0
5.0
6.3
8.0
10.0
12.5
16
20
25
32
40
50
63
80
100
1251.00
1.05
1.10
1.20
1.25
1.3
1.6
1.7
1.8
1.9
2.0
2.1
2.2
2.8
3.6
4.5
5.6
7.1
9.0
11
14
18
22
28
36
45
56
71
90
110
1401.00
1.05
1.10
1.20
1.25
1.3
1.8
1.9
2.0
2.1
2.2
2.4
2.5
3.2
4.0
5.0
6.3
8.0
10.0
12.5
16
20
25
32
40
50
63
80
100
125
160
允许线性内插值法。
a电压超过1000V,对同样材料组别,在印制线路板材料上的爬电距离和在其他绝缘材料上的爬电距离是相同的；
b对于电压超过630V,污染等级3,不推荐使用材料组别Ⅲb；
A.2.4固体绝缘
A.2.4.1概述
在设备预期寿命期间,在所有额定环境条件（见第4章）下，二次电路的固体绝缘应能承受可能发生GB4793-××××
42在正常使用时的电应力和机械应力。
如果适用，固体绝缘还需满足以下要求：
1)对模制和模压零部件，按A.2.4.2的要求；
2)对印制线路板的内绝缘层，按A.2.4.3的要求；
3)对薄膜绝缘，按A.2.4.4的要求。
A.2.4.2模制和模压零部件
对于基本绝缘、附加绝缘和加强绝缘，位于两个相同层之间的导体隔离(见图A.1，L)不小于表A.13
中规定的适用的最小距离。
表A.13距离或厚度最小值(见A.2.4.2～A.2.4.4)
交流峰值电压或直流
工作电压或重复峰值电
压最小值交流峰值电压或直流
工作电压或重复峰值
电压最小值
kV mm kV mm
＞0.0467～≤0.33 0.05 ＞8.0～≤10 3.5
＞0.33～≤0.8 0.1 ＞10～≤12 4.5
＞0.8～≤1.0 0.15 ＞12～≤15 5.5
＞1.0～≤1.2 0.2 ＞15～≤20 8
＞1.2～≤1.5 0.3 ＞20～≤25 10
＞1.5～≤2.0 0.45 ＞25～≤30 12.5
＞2.0～≤2.5 0.6 ＞30～≤40 17
＞2.5～≤3.0 0.8 ＞40～≤50 22
＞3.0～≤4.0 1.2 ＞50～≤60 27
＞4.0～≤5.0 1.5 ＞60～≤80 35
＞5.0～≤6.0 2 ＞80～≤100 45
＞6.0～≤8.0 3
A.2.4.3印制线路板内绝缘层
对基本绝缘、附加绝缘和加强绝缘,位于两个相同层之间的导体隔离不小于表A.13中规定的适用的
最小距离(见图A.2，L)。
印制线路板内绝缘层的加强绝缘，对各个单独的层也应有足够的介电强度。应使用下列方法之一:
a)绝缘厚度至少是表A.13规定的最小距离。
b)绝缘至少由印制线路板材料的两个分离的层组成，任一层的介电强度值都会被材料制造商规定
成不小于表A.9～表A.11中基本绝缘的试验电压值。
c)绝缘至少由印制线路板材料的两个分离的层组成,由材料制造商规定的组合层的介电强度至少
满足表A.9、表A.10或表A.11试验电压的1.6倍。
A.2.4.4薄膜绝缘
对于基本绝缘、附加绝缘和加强绝缘，位于两个相同层（见图A.3，L）之间的导体被隔离开，间
距应至少是A.2.2和A.2.3中的适用的电气间隙和爬电距离。
通过薄膜绝缘层的加强绝缘应有足够的介电强度。可以使用下述方法之一：
a)绝缘厚度至少应为表A.13的适用数值。GB4793-××××
43b)绝缘至少由薄膜材料的两个分离的层组成，任一层的介电强度值都会被材料制造商规定成不小
于表A.9、表A.10或表A.11中的基本绝缘的试验电压值。
c)绝缘至少由薄膜材料的三个独立层组成，任意两层都要通过试验来证明拥有合适的介电强度。
A.3条款6.7、A.1或条款A.2中未涉及的电路绝缘
A.3.1概述
这些电路具有一个或多个以下特性：
a)最大的可能瞬态过电压被电源或者在设备（见第A.4章）内限制在可知的值，该值低于假定的
电网电源电路的值；
b)最大的瞬态过电压高于假定的电网电源电路的值；
c)工作电压是多个电路的电压之和或者是混合电压；
d)工作电压包含重复峰值电压，这个重复峰值电压可能包含周期非正弦波形或者有规律的非周期
波形；
e)工作电压的频率高于30kHz。
如果是a)～c)，基本绝缘和附加绝缘的电气间隙按A.3.2确定。
如果是d)和e)，电气间隙按A.3.3确定。
对所有情况，A.3.4规定了爬电距离而A.3.5规定了固体绝缘。
A.3.2电气间隙计算
基本绝缘和附加绝缘的电气间隙按公式（A.1）确定：
电气间隙=D1+F(D2–D1)…………………………………………(A.1)
式中：
F—系数，根据以下公式（A.2）、（A.3）之一确定；
D1和D2—从表A.14中为Um取的值，
其中：
D1代表适合形状为1.250s脉冲的瞬态过电压的可能电气间隙；
D2代表适合没有任何瞬态过电压的峰值工作电压的可能电气间隙；
加强绝缘的电气间隙是基本绝缘电气间隙值的2倍。
F=(1.25Uw/Um)–0.25如果Uw/Um＞0.2...................................(A.2)
F=0 如果Uw/Um≤0.2...........................................(A.3)
式（A.2）和（A.3）中：
Um=Uw+Ut；
Uw=工作电压最大峰值；
Ut=最大附加瞬态过电压。
如果设备额定工作海拔高度高于2000m，电气间隙应乘以表A.1中的适当系数。
对于基本绝缘、附加绝缘和加强绝缘，污染等级2时最小电气间隙为0.2mm，污染等级3时最小电气间
隙为0.8mm。
表A.14A.3.2计算的电气间隙值GB4793-××××
44最大电压
Um电气间隙 最大电压
Um电气间隙
D1 D2 D1 D2
V mm mm V mm mm
14.1～266
283
330
354
453
500
566
707
800
891
1130
1410
1500
1770
2260
2500
2830
35400.010
0.010
0.010
0.013
0.027
0.036
0.052
0.081
0.099
0.12
0.19
0.38
0.45
0.75
1.25
1.45
1.74
2.440.010
0.013
0.020
0.025
0.052
0.071
0.10
0.20
0.29
0.41
0.83
1.27
1.40
1.79
2.58
3.00
3.61
5.044000
4530
5660
6000
7070
8000
8910
11300
14100
17700
22600
28300
35400
45300
56600
70700
89100
1000002.93
3.53
4.92
5.37
6.86
8.25
9.69
12.9
16.7
21.8
29.0
37.8
49.1
65.5
85.0
110
145
1656.05
7.29
10.1
10.8
13.1
15.2
17.2
22.8
29.5
38.5
51.2
66.7
86.7
116
150
195
255
290
允许线性内插值法。GB4793-××××
45A.3.3重复峰值电压或工作电压频率高于30kHz的电路的电气间隙
有重复峰值电压但频率不高于30kHz的电路的基本绝缘和附加绝缘的电气间隙应满足表A.15中
第二列的数值，并以重复峰值电压为索引。(重复峰值电压示例见图A.4)。
说明：
A——重复电压的峰值；
B——工作电压值。
图A.4重复峰值电压的示例
当电路的频率高于30kHz时，基本绝缘和附加绝缘的电气间隙应满足表A.15第三列中的数值，以工
作电压峰值为索引。
当电路可能受到重复峰值电压且频率高于30kHz时，基本绝缘和附加绝缘的电气间隙应满足上述要
求中较高的要求。
加强绝缘电气间隙是基本绝缘电气间隙值的2倍。
如果设备的额定工作海拔高度高于2000m，电气间隙应乘以表A.1中适用的系数。
对于基本绝缘、附加绝缘和加强绝缘，污染等级2的最小电气间隙是0.2mm，污染等级3的最小
电气间隙是0.8mm。
表A.15有重复峰值电压或工作电压的频率高于30kHz的电路中基本绝缘的电气间隙
电压
峰值电气间隙
频率不高于30kHz频率高于30kHz
V mm mm
0～330 0.01 0.02
400 0.02 0.04
500 0.04 0.07
600 0.06 0.11
800 0.13 0.26
1000 0.26 0.48
1200 0.42 0.76
1500 0.76 1.1
2000 1.27 1.8
2500 1.8 2.6
3000 2.4 3.5GB4793-××××
464000 3.8 5.7
5000 5.7 8
6000 7.9 10
8000 11 15
10000 15.2 20
12000 19 25
15000 25 32
20000 34 44
25000 44 58
30000 55 72
40000 77 100
50000 100
允许线性内插值法
A.3.4爬电距离
适用A.2.3的要求。
A.3.5固体绝缘
适用A.2.4的要求，但用表A.16的数值代替表A.9、表A.10或表A.11中适用的值。
从表A.16中确定需要的试验电压时，应采用以下步骤：
a)考虑A.3.3的要求，按照A.3.2计算理论上需要的电气间隙。最小电气间隙不适用于污染等级
2和3。
b)利用表A.16得出的理论上需要的电气间隙，确定需要的试验电压。
表A.16基于电气间隙的试验电压GB4793-××××
47所需电气间隙试验电压
1.2/50µs脉冲交流有效值
50/60Hz
mm Vpeak Vr.m.s.
0.010
0.025
0.040
0.063
0.1
0.2
0.3
0.5
1.0
1.5
2.0
2.5
3.0
4.0
4.5
5.0
6.0
8.0
10.0
12.0
15.0
20
25
30
40
50
60
80
100330
440
520
600
810
1150
1310
1550
1950
2560
3090
3600
4070
4930
5330
5720
6460
7840
9100
10600
12900
16400
19900
23300
29800
36000
42000
53700
65000230
310
370
420
500
620
710
840
1060
1390
1680
1960
2210
2680
2900
3110
3510
4260
4950
5780
7000
8980
10800
12700
16200
19600
22800
29200
35400
允许线性内插值法。GB4793-××××
48参考文献
[1]GB/T2900.73-2008电工术语接地与电击防护(IEC60050-195:1998,MOD)
[2]GB/T11021电气绝缘耐热性分级
[3]GB/T42125.1-20XX测量、控制和实验室用电气设备的安全要求第1部分：通用要求
[4]IEC60127(allparts)Miniaturefuses
[5]IEC60664-4Insulationcoordinationforequipmentwithinlow-voltagesystems–Part
4:Considerationofhigh-frequencyvoltagestress
[6]ISO13854Savetyofmachinery–Minimumgapstoavoidcrushingofpartsofthehumanbody
[7]ANSI/UL248-14Low-VoltageFuses–Part14:SupplementalFuses
[8]EN349Safetyofmachinery.Minimumgapstoavoidcrushingofpartsofthehumanbody
_________________________________