ICS13.310
CCSA91
GB12664—20XX
代替GB12664—2003
便携式X射线安全检查设备通用技术要求
GeneraltechnicalrequirementsforportableX-raysecurityinspection
equipment
（征求意见稿）
20XX-XX-XX发布 20XX-XX-XX实施
GB12664—20XX
I目次
前言.........................................................................................................................................................................................V
1范围..............................................................................................................................................................................................1
2规范性引用文件.......................................................................................................................................................................1
3术语和定义................................................................................................................................................................................1
4设备分类.....................................................................................................................................................................................2
5技术要求.....................................................................................................................................................................................3
5.1外观和结构.......................................................................................................................................................................3
5.2运行控制............................................................................................................................................................................4
5.3电源适用范围..................................................................................................................................................................4
5.4防电击................................................................................................................................................................................4
5.5温度限值和耐热..............................................................................................................................................................4
5.6辐射防护要求..................................................................................................................................................................4
5.7功能要求............................................................................................................................................................................5
5.8性能要求............................................................................................................................................................................6
5.9X射线产生装置安全......................................................................................................................................................7
5.10环境适应性.....................................................................................................................................................................8
5.11电磁兼容性.....................................................................................................................................................................9
6试验方法.....................................................................................................................................................................................9
6.1环境条件............................................................................................................................................................................9
6.2试验用主要仪器和工具................................................................................................................................................9
6.3外观和结构试验...........................................................................................................................................................10
6.4运行控制试验................................................................................................................................................................10
6.5电源电压适应性试验..................................................................................................................................................11
6.6防电击测试....................................................................................................................................................................11
6.7温度限值和耐热测试..................................................................................................................................................11
6.8辐射防护试验................................................................................................................................................................11
6.9功能检验试验................................................................................................................................................................12GB12664—20XX
II6.10性能指标测试.............................................................................................................................................................12
6.11环境适应性测试.........................................................................................................................................................14
6.12电磁兼容性测试.........................................................................................................................................................14
7检验规则..................................................................................................................................................................................14
7.1检验分类.........................................................................................................................................................................14
7.2型式检验.........................................................................................................................................................................14
7.3出厂检验.........................................................................................................................................................................15
7.4检验项目.........................................................................................................................................................................15
7.5判定规则.........................................................................................................................................................................15
8包装、标志、贮存和运输..................................................................................................................................................15
8.1包装..................................................................................................................................................................................15
8.2标志..................................................................................................................................................................................15
8.3贮存..................................................................................................................................................................................16
8.4运输..................................................................................................................................................................................17
9随机文件..................................................................................................................................................................................17
9.1概述..................................................................................................................................................................................17
9.2使用说明书....................................................................................................................................................................17
9.3技术说明书....................................................................................................................................................................17
附录A图像指标测试体..................................................................................................................................................18
A.1穿透力测试体................................................................................................................................................................18
A.2线分辨力测试体...........................................................................................................................................................25
A.3空间分辨力测试卡.......................................................................................................................................................27
A.4穿透分辨力测试体.......................................................................................................................................................28
A.5材料分辨与无机物分辨测试体...............................................................................................................................31
A.6有机物与灰度（混合物）分辨体...........................................................................................................................36
A.7边框测试物.....................................................................................................................................................................38
附录B周围剂量当量率测试方法................................................................................................................................40
附图GB12664—20XX
III图A.1穿透力测试体A..........................................................................................................................................................18
图A.2钢阶梯与铅条...............................................................................................................................................................19
图A.3钢阶梯.............................................................................................................................................................................19
图A.41mm厚度铅条..............................................................................................................................................................20
图A.51mm厚度钢板..............................................................................................................................................................20
图A.62mm厚度钢板..............................................................................................................................................................21
图A.75mm厚度钢板..............................................................................................................................................................21
图A.810mm厚度钢板............................................................................................................................................................22
图A.920mm厚度钢板与铅条..............................................................................................................................................23
图A.1020mm厚度钢板..........................................................................................................................................................23
图A.1110mm厚度铅条..........................................................................................................................................................24
图A.12短螺杆..........................................................................................................................................................................24
图A.13长螺杆..........................................................................................................................................................................24
图B.1线分辨力测试体..........................................................................................................................................................25
图B.2实芯铜线和薄膜装配图............................................................................................................................................26
图B.3薄膜..................................................................................................................................................................................26
图C.1空间分辨力测试卡......................................................................................................................................................27
图D.1穿透分辨力测试体.....................................................................................................................................................28
图D.2铅字和测试体固定板.................................................................................................................................................29
图D.3测试体固定板...............................................................................................................................................................29
图D.4合金铝阶梯...................................................................................................................................................................30
图D.5支架1..............................................................................................................................................................................30
图D.6支架2..............................................................................................................................................................................31
图D.7支架横梁........................................................................................................................................................................31
图E.1材料分辨与无机物分辨测试体...............................................................................................................................32
图E.2测试物固定板...............................................................................................................................................................33
图E.3尼龙..................................................................................................................................................................................33
图E.4模拟物.............................................................................................................................................................................34
图E.5PVC...................................................................................................................................................................................34
图E.6钢板..................................................................................................................................................................................35
图F.1无机物分辨测试物......................................................................................................................................................35
图G.1有机物与灰度分辨体.................................................................................................................................................36
图G.2测试物固定板...............................................................................................................................................................36
图H.1灰度分辨测试................................................................................................................................................................37
图I.1有机物分辨测试物.......................................................................................................................................................37
图J.1边框测试物.....................................................................................................................................................................38
图J.230颗直径为1mm钢珠装配图..................................................................................................................................38
图J.3有机玻璃剖面图............................................................................................................................................................39
图J.4有机玻璃沉孔放大图..................................................................................................................................................39GB12664—20XX
IV附表
表1气候环境要求..............................................................................................................................................8
表2机械环境要求..............................................................................................................................................8
表3电源适用范围试验要求............................................................................................................................11
表4不同检验类型下的检验项目...................................................................................................................15
表5线对分辨率................................................................................................................................................27
表B.1测试记录表1—应用指标测试.............................................................................................................40
表B.2测试记录表2—应用指标测试（适用于多能谱设备）.....................................................................41
表B.3测试记录表3—典型指标记录.............................................................................................................41GB12664—20XX
V前言
本文件按照GB/T1.1—2020《标准化工作导则第1部分：标准化文件的结构和起草规则》的规定
起草。
本文件代替GB12664-2003《便携式X射线安全检查设备通用规范》，与GB12664-2003相比，除结
构调整和编辑性改动外，主要技术变化如下：
——增加了5个术语：测试体、X射线产生装置、应用指标测试、典型指标测试、辐射野；
——修改了1个术语的定义：便携式X射线安全检查设备（见3.1，2003版的3.1）
——增加了设备分类；
——修改了范围（见第一章）；
——增加了设备分类（见第四章）；
——修改了外观要求（见5.1.1,2003版的4.1）
——增加了有效成像区域（见5.1.8）；
——修改了远程控制（见5.2.2,2003版的4.3.3）
——修改了电源适用范围（见5.3,2003版的4.5）
——删除了性能要求（见2003版4.8）
——增加了功能要求（见5.7）
——修改了工作状态指示（见5.7.3,2003版的4.9）
——增加了图像处理（见5.7.4.2）
——修改了性能要求（见5.8,2003版的4.8）
——修改了辐射防护要求（见5.6,2003版的4.7）
——增加了X射线产生装置安全（见5.9）
——修改了环境适应性（见5.10,2003版的4.10）
——修改了电磁兼容性（见5.11,2003版的4.11）
请注意本文件的某些内容可能涉及专利。本文件的发布机构不承担识别专利的责任。
本文件由中华人民共和国公安部提出并归口。
本文件的历次版本发布情况为：
——2003年首次发布为GB12664-2003；
——本次为第一次修订。GB12664—20XX
1便携式X射线安全检查设备通用技术要求
1范围
本文件规定了便携式X射线安全检查设备的分类、技术要求、试验方法、检验规则、包装、标志、
贮存和运输以及随机技术文件等要求，描述了试验方法，确立了检验规则。
本文件适用于各种便携式X射线安全检查设备的设计、制造和检验。
本文件不适用于基于X射线背散射技术的便携式X射线安全检查设备。
2规范性引用文件
下列文件中的内容通过文中的规范性引用而构成本文件必不可少的条款。其中，注日期的引用文件，
仅该日期对应的版本适用于本文件。不注日期的引用文件，其最新版本（包括所有的修改单）适用于本
文件。
GB/T191—2008包装储运图示标志
GB/T2423.1—2008电工电子产品环境试验第2部分：试验方法试验A：低温
GB/T2423.2—2008电工电子产品环境试验第2部分：试验方法试验B：高温
GB/T2423.3—2016环境试验第2部分：试验方法试验Cab：恒定湿热试验
GB/T2423.5—2019电工电子产品环境试验第2部分：试验方法试验Ea和导则：冲击
GB/T2423.10—2019电工电子产品环境试验第2部分：试验方法试验Fc:振动（正弦）
GB/T4208—2017外壳防护等级（IP代码）
GB4793.1—2007测量、控制和实验室用电气设备的安全要求第1部分：通用要求
GB/T4835.1—2012辐射防护仪器β、X和γ辐射周围和/或定向剂量当量（率）仪和/或监测仪第
1部分：便携式工作场所和环境测量仪与监测仪
GB4943.1—2011信息技术设备安全第1部分：通用要求
GB15208.1—2018微剂量X射线安全检查设备第1部分：通用技术要求
GB15208.2—2018微剂量X射线安全检查设备第2部分：透射式行包安全检查设备
GB16796—2009安全防范报警设备安全要求和试验方法
GB/T17626.2—2018电磁兼容试验和测量技术静电放电抗扰度试验
GB/T17626.3—2016电磁兼容试验和测量技术射频电磁场辐射抗扰度试验
GB/T17626.4—2018电磁兼容试验和测量技术电快速瞬变脉冲群抗扰度试验
GB/T17626.5—2019电磁兼容试验和测量技术浪涌（冲击）抗扰度试验
GB/T17626.6—2017电磁兼容试验和测量技术射频场感应的传导骚扰抗扰度试验
GB18871—2002电离辐射防护与辐射源安全基本标准
3术语和定义
GB15208界定的以及下列术语和定义适用于本文件。GB12664—20XX
23.1
便携式X射线安全检查设备portableX-raysecurityinspectionequipment
一种利用X射线的穿透特性进行现场安全检查的装置，其结构整体或分成单元一般可由工作人员直
接携带。
注：在本文件中便携式X射线安全检查设备简称设备；可以单独携带的设备的一部分简称单元；安装或依附在单元
上功能组件简称部件。
3.2
工作位置operationposition
按设备的使用要求正常操作设备时，工作人员相对于X射线产生装置的位置。
3.3
测试体testobject
用于测试和评价X射线图像性能指标的一组测试物。
3.4
X射线产生装置X-raygeneratingdevice
产生和控制X射线发射所有部件的组合。
注：通常包括X射线管、高压产生器、控制器以及冷却系统组成。
3.5
应用指标测试Comprehensivetest
被测对象紧贴成像装置。保持X射线产生装置、成像装置位置不变，X射线束能够覆盖全部有效成像
区域，通过调整X射线产生装置、成像装置的参数，对设备各项性能指标的最优值进行的测试。
3.6
典型指标测试Optimaltest
被测对象紧贴成像装置。通过调整X射线产生装置、成像装置的位置和参数，对设备各项性能指标
的最优值进行的测试。
3.7
辐射野radiationfieldsize
与X射线束相交的平面内,其中辐射强度超过某一比例或规定水平的区域。
4设备分类
4.1便携式X射线安全检查设备（以下简称设备）按照使用环境分为：
a)室内型设备；
b)室外型设备。GB12664—20XX
34.2设备按照材料分辨能力分为：
a)单能谱型设备；
b)多能谱型设备。
5技术要求
5.1外观和结构
5.1.1外观
a)设备的外观应完好，表面应平整光洁，不应有凹陷、裂纹、划痕等明显的机械损伤或永久变形；
b)涂镀层光洁，不应有起泡、剥落、腐蚀、污渍等缺陷；金属件应无锈蚀、塑料件应无起泡、开
裂；
c)常触及的部位不应有可能勾扯衣物或划伤皮肤的钉头、尖角、毛边或锐棱；
d)设备的开/关机按键和其他控制部件的操作应灵活可靠，布局应方便实用，开关机按键应具有
防误操作功能。设计及操作程序应符合人体工效学的基本要求，并便于操作和维修。
5.1.2设备结构
设备包括的部件和所有零件，应有足够的机械强度和刚度，工作状态下应能放置平稳，无晃动；所
有调节和控制机构应安装正确、操作灵活。
5.1.3外壳防护等级
设备外壳防护等级应不低于GB/T4208—2017中5.1规定的IP50的要求。
5.1.4提手
携带单元的提手及其与主体结构的连接应足够牢固，不应有断裂、变形或其他明显失效现象。
5.1.5脚轮
具有脚轮的携带单元，其脚轮应有足够的强度和转动灵活性，与设备的连接应牢固可靠。
5.1.6连接电缆
要求如下：
a)电缆的收放、接插和分断应设计合理，便于操作；
b)不同用途的插头之间应不能互换插接；
c)电缆根部与接插件的连接须足够牢固，应有防止过度弯折的措施，并至少应能承受50N的拉力
而不出现松脱现象。
5.1.7密封部件
X射线产生装置的密封部件应保持良好的性能，无渗漏现象。
5.1.8有效成像区域
成像装置的面板应明确标识出可成像的有效区域。
使用手册中应明确说明可成像的区域及尺寸，不可成像的边框尺寸。GB12664—20XX
45.2运行控制
5.2.1控制装置的操作
要求如下：
a)开关、按钮和各种装置应操作灵活，手感明确，功能可靠；
b)如果设备处在无线工作模式下，工作距离至少应达到10m；
c)电池供电设备的电源应由防触碰开关控制，以防止非工作状态时误通电。
5.2.2远程控制
如果通过远程计算机或网络进行控制，则应提供相应的控制程序。当远程控制因故中断时，设备应
能自动停机或恢复本地控制。
5.3电源适用范围
如果设备具备交流供电和电池供电能力，应满足5.3.1、5.3.2的要求。如只具备交流供电能力，
应满足5.3.1要求。如只具备电池供电能力，应满足5.3.2要求。
5.3.1交流供电
要求如下：
a)适用交流供电的设备，电源电压在标称电压的+10%
-15%和标称频率的±3Hz范围内，设备应能正常
工作；
b)对电源有特殊要求的单元或部件，应在相应的使用说明中明示。
5.3.2电池供电
要求如下：
a)设备应配置可充电电池。在电池充满或电池电压超出正常工作范围时，设备应能告知工作人员
注意；
b)使用可更换电池供电的设备，应具有防止电池反接的设计；
c)电池不得由于过度充电、放电或由于电池安装时极性不正确而引起爆炸或出现着火危险，应符
合GB4793.1-2007中13.2.2的要求；
d)可充电电池应能保证设备在正常状态下，应至少连续采集100幅透视图像；
e)采用锂电池供电的设备，其使用的全部锂电池应具备3C认证或符合GB31241-2014的要求。
5.4防电击
设备在正常条件和单一故障条件下均应当保持防电击，设备的可触及零部件不应出现危险带电，符
合GB4793.1-2007中第6章的要求。
5.5温度限值和耐热
设备的温度限制和耐热应符合GB4793.1-2007中第10章的要求。
5.6辐射防护要求
5.6.1辐射控制装置GB12664—20XX
5要求如下：
a)启动辐射和停止辐射时需要操作的控制装置必须置于X射线辐射野之外；
b)利用距离防护的上述控制装置，其连接到X射线产生装置控制端的电缆长度不应短于10m。如
果使用延时控制装置实现距离防护，延迟时间不应小于10s。
5.6.2泄漏辐射
距X射线辐射窗口1m处，周围剂量当量率应不超过1mSv/h。
5.6.3杂散辐射
要求如下：
a)当X射线出束时，应在其周围设置控制区，控制区边界每小时累积周围剂量当量应不超过
20μSv；
b)当X射线出束时，应在其周围设置监督区，公众禁止进入该区域，监督区边界每小时累积周围
剂量当量应不超过2.5μSv；
c)操作人员位置，每小时累积周围剂量当量应不超过0.5μSv；
d)使用说明中，应提供设备、控制区、监督区和设备操作人员的距离、位置说明。
注：如因工作环境所限、操作人员位置剂量超出上述限值时，应由按GBZ128的要求进行个人剂量监测、按GBZ98
的要求进行职业健康监护的人员操作，操作人员应佩戴个人剂量计、个人剂量报警仪。
5.6.4辐射野的限制
X射线辐射窗口上应安装限束系统（如限束器、光阑等），保证辐射野外的辐射强度有明显下降。
辐射野中心、辐射野边缘外的测量值，两者至少应相差5倍。
5.7功能要求
5.7.1功能
设备在正常条件下进行操作，应能完成使用说明中规定的各项功能。
5.7.2工作状态指示
要求如下：
a)设备应有明显的系统工作和射线发射显示装置（指示灯）；
b)设备电源的接通应有明显指示；
c)X射线出束时应有明显灯光指示，且指示灯应为红色，且在X射线发射过程中保持亮起，最小
显示时间不应短于X射线出束时间；
d)X射线出束时应有明显声音提示，声级应不低于85dB(A)，且在X射线发射过程中持续发声，
最小持续时间不应短于5.0s；
e)如果有曝光延时的计时器被激活并在发射倒计时的状态，X射线出束与倒计时状态的声音报警
和视觉报警应可以区分；
f)设备发生故障时应有明确指示；
g)在6000lx的明亮光线和25lx的昏暗光线下，均应能清晰地分辨出各种指示；
h)使用电池作为电源的设备及部件，应有电池状态的指示（如：正常、满电、欠压或故障等）；
i)操作面板和屏幕文本信息应为图标或中文界面。GB12664—20XX
65.7.3图像
5.7.3.1图像显示
满足以下要求：
a)X射线图像能完整显示被检对象及内部物体的整体轮廓；
b)设备应能通过移动成像装置等方式，通过多次拍摄，将拍摄的图像进行拼接，形成完整的被检
对象透视图像。
5.7.3.2图像处理
满足以下要求：
a)图像增强功能：增强显示图像局部细节；
b)有机物和无机物剔除（多能谱设备适用）；
——有机物剔除时，剔除图像中的等效有机物，显示剩余物质的颜色；
——无机物剔除时，剔除图像中的等效无机物，显示剩余物质的颜色；
c)图像放大功能：放大显示所选中区域的物体图像，任意区域放大应不小于4倍；
d)图像比较功能：至少应能同时显示并对比两幅图像；
e)设备应对穿不透区域给予明显提示。
5.7.3.3图像存储
图像存储要求如下：
a)图像存储功能：图像应自动存储；
b)图像存储容量：可存储不少于10000幅被检图像；
c)图像存储格式：
——应能通过生产厂家专用软件再现图像，图像处理功能与设备功能相同；
——应能将图像的原始数据转换成通用的图像格式。
5.7.3.4帧率
具备动态成像功能的设备，帧率应满足使用说明中的要求，单位是fps。
5.7.4设备自诊断功能
设备应具备自诊断功能，并生成至少包括X射线产生装置、X射线探测装置、控制器等主要功能部件
运行状态的诊断报告。
5.7.5射线装置开关
X射线产生装置至少应具有防触碰开关，开关应能清楚地识别“通”“断”位置。
5.8性能要求
5.8.1概述
应保持测试体平面垂直于射线发射方向，允许采用图像处理功能取得最佳评价效果。
应用指标应满足5.8.2-5.8.10的要求，典型指标应不低于应用指标，且满足使用说明中的要求。GB12664—20XX
75.8.2穿透力
应用指标测试中，静态成像设备应能穿透厚度为6mm的钢板；具备动态成像功能的设备，在动态成
像工作模式下应能穿透厚度为3mm的钢板。
5.8.3线分辨力
应用指标测试中，设备应能分辨直径为0.2mm的单根铜线。
5.8.4空间分辨力
应用指标测试中，静态成像设备应能分辨1.2lp/mm；
具备动态成像功能的设备，在动态成像最大帧频模式下应能分辨0.6lp/mm。
5.8.5穿透分辨力
应用指标测试中，静态成像设备应能分辨厚度为9.5mm、15.9mm、22.2mm合金铝阶梯下，直径为
0.5mm的单根实心铜线；
具备动态成像功能的设备，在动态成像最大帧频模式下应能应能分辨厚度为9.5mm、15.9mm、22.2mm
合金铝阶梯下，直径为1mm的单根实心铜线。
5.8.6灰度分辨
应用指标测试中，静态成像设备应能分辨厚度范围为1mm～40mm，厚度差不小于2mm的合金铝阶梯。
具备动态成像功能的设备，在动态成像工作模式下应能分辨厚度范围为1mm～20mm，厚度差不小于
2mm的合金铝阶梯。
5.8.7材料分辨（多能谱型设备适用）
应能分辨具有不同等效原子序数的四种材料样本，并赋予PVC板绿色，赋予模拟物板和尼龙6板橙
色，赋予钢板蓝色。
5.8.8有机物分辨（多能谱型设备适用）
使用说明中应规定设备有机物最大识别厚度，用有机物阶梯厚度表示。即能分辨并赋予该厚度及以
下有机物阶梯不同饱和度、且均匀的橙色。
5.8.9混合物分辨（多能谱型设备适用）
使用说明中应规定设备的混合物最大识别厚度，用合金铝阶梯厚度表示。即能分辨并赋予该厚度及
以下合金铝阶梯不同饱和度、且均匀的绿色。
5.8.10无机物分辨（多能谱型设备适用）
使用说明中应规定设备的无机物最大识别厚度，用钢阶梯厚度表示。即能分辨并赋予该厚度及以下
钢阶梯不同饱和度、且均匀的蓝色。
5.9X射线产生装置安全
X射线产生装置应在设备内实现自冷却；
X射线产生装置应具有过电压和过电流保护功能，当其电压或电流超过产品规定值时，能自动切断
高压。GB12664—20XX
85.10环境适应性
5.10.1气候环境适应性
按表1进行气候环境适应性试验，试验过程中不发生状态改变，试验后设备应能正常工作。
表1气候环境要求
试验项目严酷等级
试验方法整机 具有独立功能的电器部件
室内型
设备室外型设
备持续时间
H检查项目持续时间
h检查项目
低温（5±2）
℃（-15±2
）℃按GB/T2423.1—2008试
验Ab进行，试验过程通电。2试验开始
前的初试
试验和每
项试验结
束前的检
查项目和
方法由使
用说明规
定2
试验开始前
的初试试验
和每项试验
结束前的检
查项目和方
法由使用说
明规定高温（35±2
）℃（50±2）
℃按GB/T2423.2—2008试
验Bb进行，试验过程通电。22
恒定湿热（40±2）℃
相对湿度：93±3%
（不结露）按GB/T2423.3—2016试
验Cab进行，试验过程通
电。16 16
低温存贮 -15±3℃按GB/T2423.1—2008试
验Ab进行，试验过程不通
电。16试验结束
后至少恢
复2h后检
测，检查
项目和方
法由使用
说明规定16试验结束后
至少恢复2h
后检测，检
查项目和方
法由使用说
明规定高温存贮 50±3℃按GB/T2423.2—2008试
验Bb进行，试验过程不通
电。16 16
注：超越正常环境条件的性能测试，允许另行规定或调整参数。
5.10.2机械环境适应性
按表2的规定进行机械环境适应性试验，试验后设备应能正常工作，各部件不应产生永久性的结构
变形、机械损伤、电气故障和紧固件松动，内部线路、电路板和接口等接插件不应有脱落、松动或解除
不良现象，存储数据不应丢失。
表2机械环境要求
试验项目 严酷等级设备状态
振动试验
Fc频率范围/Hz：10～55～10（正弦波）
振幅/mm：0.15
振动方向：X、Y、Z
持续时间/min：45裸机，工作状态
冲击试验
Ea峰值加速度/（m/s2）：150
持续时间/ms：11裸机，工作状态GB12664—20XX
9冲击方向：6个轴向
冲击次数：每个轴向3次，共18次
自由跌落（含包装
箱）跌落高度：250mm
地面：水泥或瓷砖地面
实验次数：6（包装箱每面一次）含包装箱、非工作状态
注：对于不宜与试验设备台面直接接触的液晶显示器屏幕、平板探测器探测面等部位，可夹不超过5mm的缓冲衬垫。
5.11电磁兼容性
对设备进行静电放电、射频电磁场辐射、电快速瞬变脉冲群、浪涌(冲击)、射频场感应的传导骚扰
共5项抗扰度试验,抗扰度性能应符合GB/T17799.1—2017中表1、表2和表4中规定限值的要求。
6试验方法
6.1环境条件
除另有规定外，全部试验环境条件均为正常大气条件：
——环境温度：15℃～35℃；
——相对湿度：15％～75％；
——大气压力：86kPa～106kPa；
——附近没有足以影响测试结果的其他电离辐射源工作；
——以X射线出束窗口为圆心，半径2m的范围内不应有足以影响测试结果的X射线散射体；
——设备处于使用说明规定的默认工作参数。
6.2试验用主要仪器和工具
6.2.1辐射防护测试用剂量仪
应满足如下要求：
a)仪表宜选用直接测量周围剂量当量率的测量仪表，单位为希沃特每小时（Sv/h）；
b)读数分辨力小于或等于0.01uSv/h；
c)能量响应在50-500KeV范围内不得超过±30%；
d)相对固有误差：在有效量程内不得超过±15%；
e)过载指示：当剂量率大于量程最大值时，仪表应指示值大于满刻度或给出过载的指示；
f)仪表积累剂量测量的各个档位应按照国家计量检定规程定期进行计量校准；
g)脉冲式X射线产生装置的测试应选用响应时间为20纳秒（或以内）的辐射剂量率测量仪表。
6.2.2声级计
频率范围25Hz～8kHz，采用A计权声级模式。应按照国家计量检定规程定期进行计量校准。
6.2.3秒表
误差小于或等于0.01s。
6.2.4卷尺
按照国家计量检定规程定期进行计量校准。GB12664—20XX
106.2.5电子秤
按照国家计量检定规程定期进行计量校准。
6.3外观和结构试验
6.3.1外观检查
按5.1.1的要求进行检查，采用实物与设计文件核对、观察及手动等方式进行，判断结果是否符合
要求。
6.3.2结构检查
采用实物与设计文件核对，观察及触摸等人工方法对设备进行检查，判定结果是否符合5.1.2的要
求。
注：结构设计文件一般指设备机械设计图。
6.3.3外壳防护等级测试
按GB/T4208—2017中第13章对外壳防护等级进行试验，判定结果是否符合5.1.3的要求。
6.3.4提手
按GB4793.1—2007的7.4进行提手牢固度试验，应符合5.1.4的要求。
6.3.5脚轮
按GB4793.1—2007的8.4对装有脚轮的底边进行跌落试验，应符合5.1.5的要求。
6.3.6连接电缆
固定电缆插头，用50N的拉力牵拉电缆线5分钟，然后检查电缆与接插件的连接和电缆在设备上的
插接，均应符合5.1.6的要求。
6.3.7密封部件
目视或由使用说明规定方法进行试验，应符合5.1.7的要求。
6.3.8有效成像区域
将测试体I贴近放置在成像装置边框处，启动并完成扫描成像，目测透视图像中的钢珠数量。根据
钢珠数量，计算出不可成像的上、下、左、右边框宽度并记录；
使用卷尺测量成像装置尺寸，计算出有效成像区域面积并记录。
注：每颗钢珠直径1mm。
6.4运行控制试验
6.4.1控制装置操作
目视检查并按照使用说明操作，应符合5.2.1的要求。
6.4.2远程控制
按照使用说明操作，应符合5.2.2的要求。GB12664—20XX
116.5电源电压适应性试验
6.5.1交流供电试验
使用交流供电的设备，电源使用范围试验要求见表3，在表3规定的9个试验点各试验15min，每
个试验点采集5次透视图像，判定设备是否能正常工作。
表3电源适用范围试验要求
序号 交流电压 频率
1 标称值 标称值
2 标称值 标称值+3Hz
3 标称值 标称值-3Hz
4 标称值的85% 标称值
5 标称值的85% 标称值+3Hz
6 标称值的85% 标称值-3Hz
7 标称值的110% 标称值
8 标称值的110% 标称值+3Hz
9 标称值的110% 标称值-3Hz
6.5.2电池供电试验
要求如下：
a)使用电池供电的设备，目视检查并按照使用说明操作；
b)使用厂商推荐的充满电的电池供电。开启设备，从设备进入工作状态开始，参数按5.8.2测试
要求配置，测试过程中应至少进行100次透视图像采集，观察设备状态和是否出现电量不足或
欠压提示，设备状态异常和出现电量不足或欠压提示均判定电源电量不足。判定测试结果是否
符合5.3.2的要求。
6.6防电击测试
按GB4793.1-2007中第6章的规定的试验方法进行，判定结果是否符合5.4的要求。
6.7温度限值和耐热测试
按GB4793.1-2007中第10章的规定的试验方法进行，判定结果是否符合5.5的要求。
6.8辐射防护试验
进行本项试验时，应记录X射线产生装置的重量。
6.8.1辐射控制装置
按使用说明规定进行操作、核验和测量应符合5.6.1的要求。
6.8.2泄漏辐射试验
X射线束出口应当用不小于10mm厚度的铅块屏蔽好，然后用辐射仪按以下方法测量泄漏剂量率：
a)将X射线产生装置设置在最高工作管电压和在该电压对应的最大束流下；GB12664—20XX
12b)以X射线束出射窗口为圆心、半径1m的平面上至少选5点进行测量，5个点的泄漏辐射均应符
合5.6.2的要求；
6.8.3杂散辐射试验
a)按附录C规定的试验方法对设备进行试验，判定结果是否符合5.2.2的要求。
6.8.4辐射野的限制试验
用剂量仪按以下方法获得的辐射野中心、辐射野边缘外的测量值，两者至少应相差5倍。
a)设备按工作状态放置就位，并按产品有关规定安装限束系统；
b)在成像装置与辐射束轴线的交点处测量辐射野中心的剂量当量；
c)在成像装置平面上，距离平面中心点最远的辐射野边缘外侧0.1m处任选3点，测量辐射野边
缘外的剂量当量。
6.9功能检验试验
6.9.1功能检测
在正常条件下，按使用说明进行功能操作，应符合5.7.1的要求。
6.9.2工作状态指示检验
目视检查和操作设备，应符合5.7.3的要求。
在无阳光直射和反光的情况下，分别将设备附近的环境光照度提高到6000lx±600lx以及降低到
25lx±3lx，并在正常操作位置观察各种指示，应符合5.7.3的要求。
6.9.3图像功能测试
按照使用说明操作，判定结果是否符合5.7.3的要求。
设置成像装置为最大帧频模式下，对准运行的秒表，录制不短于30秒的视频。保存后查看并计算
帧数，判定结果是否符合5.7.3.4的要求。
注：需记录成像装置的完整型号。
6.9.4设备自诊断功能测试
按照使用说明操作，判定结果是否符合5.7.5的要求。
6.9.5射线开启确认测试
按照使用说明操作，判定结果是否符合5.7.6的要求。
6.10性能指标测试
按照使用说明和5.8的要求进行试验，判断结果是否符合要求。
6.10.1概述
测试物平面应垂直于X射线发射方向以得到最佳测试图像。
应进行应用指标测试和典型指标测试。
记录格式参见附录C。
6.10.2穿透力测试GB12664—20XX
13按照以下内容进行试验：
a)将测试体A平稳放置在检查区域，启动并完成扫描成像，目测显示器上测试体A的X射线图像，
对设备进行穿透能力的测试，判定结果是否符合5.8.2的要求；
b)观察设备对穿不透区域的警示信息，判定结果是否符合5.7.4.2e)的要求。
注：如果可以看到被钢阶梯遮挡的铅条的绝大部分，则可认为设备能穿透此钢阶梯。
6.10.3线分辨力测试
将测试体B平稳放置，启动并完成扫描，目测显示器上测试体B的X射线图像，判定结果是否符合
5.8.3的要求。
注：如果可以看到金属线的绝大部分，则可认为设备能分辨此金属线。
6.10.4空间分辨力测试
将测试体C平稳放置，启动并完成扫描，目测显示器上测试体C的X射线图像，判定结果是否符合
5.8.4的要求。
注：如果金属线能完整区分开，则可认为设备能分辨此线对。
6.10.5穿透分辨力测试
将测试体D平稳放置，启动并完成扫描，目测显示器上测试体D的X射线图像，判定结果是否符合
5.8.5的要求。
注：如果可以看到被合金铝阶梯遮挡的金属线的绝大部分，则可认为设备能分辨此金属线。
6.10.6灰度分辨测试
将测试体F平稳放置，启动并完成扫描，目测显示器上测试体F中测试卡X的X射线图像，判定结
果是否符合5.8.6的要求。
注：如果可以将合金铝阶梯样本的相邻阶梯区分开，并能赋予不同的灰度，则可认为设备能分辨。
6.10.7材料分辨测试（多能谱设备适用）
将测试体E平稳放置，启动并完成扫描，目测显示器上测试体E中测试卡X的X射线图像，判定结
果是否符合5.8.7的要求。
注：如果样本图像呈现相同的灰度，不同的颜色，则可认为设备能分辨这些样本。
6.10.8有机物分辨测试（多能谱设备适用）
将测试体G平稳放置，启动并完成扫描，目测显示器上测试体F中测试卡X的X射线图像，判定结
果是否符合使用说明的要求。
注：能分辨并赋予该阶梯及以下厚度有机物阶梯不同饱和度、且均匀的橙色，则可认为设备能分辨该阶梯以下区域。
6.10.9混合物分辨测试（多能谱设备适用）
将测试体H放置在检查区域，启动并完成扫描，目测显示器上测试体F中测试卡X的X射线图像，
判定结果是否符合使用说明的规定。
注：能分辨并赋予该阶梯及以下厚度有机物阶梯不同饱和度、且均匀的绿色，则可认为设备能分辨该阶梯以下区域。
6.10.10无机物分辨测试（多能谱设备适用）GB12664—20XX
14将测试体F放置在检查区域，启动并完成扫描，目测显示器上测试体F中测试卡X的X射线图像，
判定结果是否符合使用说明的规定。
注：能分辨并赋予该阶梯及以下厚度有机物阶梯不同饱和度、且均匀的蓝色，则可认为设备能分辨该阶梯以下区域。
6.11环境适应性测试
按表1和表2的要求及GB/T2423.1—2008、GB/T2423.2—2008、GB/T2423.3-2016、GB/T
2423.5—2019和GB/T2423.10-2019规定的试验方法进行。
6.12电磁兼容性测试
6.12.1静电放电抗扰度试验
按GB/T17626.2—2018中规定的试验方法对设备进行静电放电抗扰度试验，实验等级应不低于3
级。判定结果是否符合5的要求。
6.12.2射频电磁场辐射抗扰度试验
按GB/T17626.3—2016中规定的试验方法对设备进行射频电磁辐射抗扰度试验，判定结果是否符
合5的要求。
6.12.3浪涌（冲击）抗扰度试验
按GB/T17626.5—2019中规定的试验方法对设备进行浪涌（冲击）抗扰度试验，判定结果是否符
合5的要求。
6.12.4电快速瞬变脉冲群抗扰度试验
按GB/T17626.4—2018中规定的试验方法对设备进行浪涌（冲击）抗扰度试验，判定结果是否
符合5的要求。
6.12.5射频场感应的传导骚扰试验
按GB/T17626.6—2017中规定的试验方法对设备进行浪涌（冲击）抗扰度试验，判定结果是否
符合5的要求。
7检验规则
7.1检验分类
设备检验分为型式检验和出厂检验。
7.2型式检验
7.2.1型式检验要求
当产品遇有下列情况之一时，应进行型式检验：
a)在设计定型和生产定型时；
b)产品的设计、工艺、生产设备、管理等方面有较大改变（包括人员素质的改变）而影响到产品
的性能；
c)产品长期（一年以上）停产后，恢复生产；GB12664—20XX
15d)国家质量监督机构要求进行时。
7.2.2型式检验的抽样规则
随机抽取1～2台样品进行型式检验。
7.3出厂检验
所有设备在出厂交付使用前都应逐台进行出厂检验。
7.4检验项目
检验项目见表4。
表4不同检验类型下的检验项目
序号 检验项目 技术要求 试验方法 型式检验 出厂检验
1外观和结构 5.1 6.3 ● ●
2运行控制 5.2 6.4 ● ●
3电源适用范围 5.3 6.5 ● ―
4防电击 5.4 6.6 ● ―
5温度限制和耐热 5.5 6.7 ● ―
6辐射控制装置 5.6.1 6.8.1 ● ●
7泄漏辐射 5.6.2 6.8.2 ● ●
8杂散辐射 5.6.3 6.8.3 ●
9辐射野的限制 5.6.4 6.8.4 ●
10功能要求 5.7 6.10 ● ●
11性能要求 5.8 6.11 ● ●
12环境适应性 5.10 6.12 ● ―
13电磁兼容性 5.11 6.13 ●
7.5判定规则
型式检验中如发现不合格项，允许对产品的相关部件进行不超过两次的调整或更换，并重新检验。
仍不合格，则应判为不合格品。
出厂检验中如发现不合格项，允许对产品的相关部件进行不超过两次的调整或更换，并重新检验。
仍不合格，则应判为不合格品。
8包装、标志、贮存和运输
8.1包装
设备的包装要求：
a)包装箱应能适应常用运输条件；
b)设备在包装箱内要可靠固定；
c)包装箱要防潮、防震。
8.2标志GB12664—20XX
168.2.1标识
要求如下：
a)在供电电源端子附近应标出电源的额定供电电压，或使用电池的型号和连接极性；
b)开关的通断、按钮的功能、各种现场连接的线缆以及备选件的安装均应标识清晰、明确；
c)在可更换的熔断器附近，应标明熔断器的额定值；
d)有可能引起危险的操作装置或步骤应在醒目的位置加警告标记，并在使用说明中突出注明；
e)如果产品的辐射输出可能对感光胶片等物品产生有害影响，则应在醒目的位置提示注意，并在
使用说明中说明；
f)用图形表示的符号应符合GB4793.1—2007表1的要求；
g)需要文字表示的应当用中文标出；
h)面板上的标记、字迹要清晰。在正常使用和厂家规定的清洁方法下，文字、符号和标识应能长
期保持清晰和牢固；
i)成像装置应明确标识出实际成像区域。
8.2.2设备标志
在设备的适当位置上应有下列标志：
a)产品型号、生产日期、编号和厂家；
b)标称电压、标称频率、标称电流和功率；
c)射线源的管电压、管电流；
d)设备外形尺寸（mm）：高×宽×深；
e)设备重量（kg）；
f)警告性说明应标在设备显著的位置；对设备内、外表面上的警告性说明应标在其附近，或标在
有关部件上或其附近；
8.2.3包装箱标志
在设备包装箱上应有下列标志：
a)产品型号、名称和数量；
b)箱体外形尺寸（mm）：长×宽×高；
c)装箱毛重（kg）；
d)装箱日期（年、月）；
e)易见处应有防潮、防震、严禁倒置，以及叉车插入位置等标志或字样，标志图示符合GB/T191
—2008的规定。
8.3贮存
8.3.1短期贮存
在运输或贮存包装状态下，设备应能在不超出下列范围的环境条件下放置15周以上：
a)环境温度：―40℃～60℃；
b)相对湿度：10％～90％。
8.3.2长期贮存GB12664—20XX
17需要长期存放的设备，应有良好的贮存条件，库房应清洁干燥，通风良好，周围不得有腐蚀性气体，
相对湿度不大于80％，设备应在包装箱内。
8.4运输
设备的运输要求：
a)包装好的设备可用空、海、陆交通工具运输，运输过程中应避免雨、雪的直接淋袭；
b)设备在车站、码头中转时，应存放在库房内。
9随机文件
9.1概述
随机技术文件被视为设备的组成部分。应至少包括使用说明书、技术说明书、产品合格证和装箱清
单文件。
警告性说明和警告性符号（标在设备上的）解释应在随机技术文件中给出。
9.2使用说明书
使用说明书应提供使设备按其技术条件运行的全部资料，应至少包括：
a)安装和拆卸方法；
b)基本工作原理和操作说明；
c)各部件之间的电缆连接；
d)与附件或其他设备连接的说明；
e)工作电压范围、电源频率范围和功耗；
f)工作环境和贮存环境的温湿度范围；
g)外形尺寸和重量；
h)操作控制装置的识别和使用；
i)显示和报警信息的说明；
j)日常维护、检查、保养和清洁。
9.3技术说明书
技术说明书应提供设备维护维修所需的全部技术信息，应至少包括：
a)设备组成框图；
b)主要功能及其技术要求；
c)各部件的功能描述；
d)设备工作频率范围；
e)主要部件更换和调试方法；
f)系统的机械和电气连接框图；
g)保障安全使用应注意的事项；
h)常见故障的处理；
i)供电、信号以及电缆连接图；
j)制造厂商详细名称和地址；
k)技术服务和维修部门的联络信息。GB12664—20XX
18附录A
（规范性）
图像指标测试体
A.1穿透力测试体
穿透力测试物A用于穿透力测试。其包括：钢阶梯（背粘1mm厚度铅条）、1mm厚度钢板、2mm厚度钢
板、5mm厚度钢板、10mm厚度钢板、20mm厚度钢板（背粘10mm厚度铅条）。测试时可通过螺杆组合钢板
形成所需测试厚度。
图A.1穿透力测试体AGB12664—20XX
19
说明：
①——钢阶梯；
②——厚度为1mm的铅条。
图A.2钢阶梯与铅条
注：材料为45#钢。
图A.3钢阶梯GB12664—20XX
20
注：材料为铅Pb。
图A.41mm厚度铅条
注：材料为45#钢。
图A.51mm厚度钢板GB12664—20XX
21
注：材料为45#钢。
图A.62mm厚度钢板
注：材料为45#钢。
图A.75mm厚度钢板GB12664—20XX
22
注：材料为45#钢。
图A.810mm厚度钢板GB12664—20XX
23
说明：
①——20mm厚度钢板；
②——10mm厚度铅条。
图A.920mm厚度钢板与铅条
注：材料为45#钢。
图A.1020mm厚度钢板GB12664—20XX
24
注：材料为铅Pb。
图A.1110mm厚度铅条
图A.12短螺杆
图A.13长螺杆GB12664—20XX
25A.2线分辨力测试体
线分辨力测试体B由一组正弦曲线形单根实芯铜线和薄膜组成。这些铜线用两层厚度为0.1mm的薄膜
夹紧固定，单根实芯铜线一共16根，直径从低到高分别是0.01mm、0.02mm、0.03mm、0.04mm、0.05mm、
0.06mm、0.07mm、0.08mm、0.09mm、0.10mm、0.12mm、0.14mm、0.16mm、0.20mm、0.22mm、0.24mm。
单位（mm）
0.01
0.02
0.03
0.04
0.05
0.06
0.07
0.08
0.09
0.10
0.12
0.14
0.16
0.20
0.22
0.24①
②
③
说明：
①——不同直径单根实芯铜线；
②——粘有不同直径单根实芯铜线的薄膜；
③——铜丝直径标志铜箔。
图B.1线分辨力测试体GB12664—20XX
26
图B.2实芯铜线和薄膜装配图
注：材料为薄膜。
图B.3薄膜GB12664—20XX
27A.3空间分辨力测试卡
空间分辨力测试卡C，用于测试设备分辨线对的能力。测试卡由两层透明的树脂玻璃中间夹一层
0.1mm厚的铅箔构成。空间分辨率用线对/毫米（lp/mm）为单位。
该测试卡分辨率范围：0.6-10.0（lp/mm），组数为26组,分辨率从小到大排列，最小为0.6，最大
为10，详见表5。
表5线对分辨率
图C.1空间分辨力测试卡分辨
率
（lp/
mm）0.60.70.80.91.01.21.41.61.82.02.22.52.8
3.13.43.74.04.34.65.05.56.07.08.09.010GB12664—20XX
28A.4穿透分辨力测试体
穿透分辨力测试体D由测试物固定板、铝型材支架、合金铝阶梯（5A02）、一组正弦曲线形单根实
芯铜线和铅字组成，实芯铜线固定在凹槽内，铝阶梯覆盖在其上方，铜丝直径分别为：2mm、1mm、0.5mm、
0.4mm、0.3mm、0.2mm、0.1mm、0.08mm、0.06mm、0.04mm、0.02mm。
④
说明：
①——合金铝阶梯；
②——铝型材支架；
③——测试体固定板；
④——铅字。
图D.1穿透分辨力测试体GB12664—20XX
29
图D.2铅字和测试体固定板
注：材料为ABS。
图D.3测试体固定板GB12664—20XX
30
注：材料为5A02。
图D.4合金铝阶梯
注：材料为5A02。
图D.5支架1GB12664—20XX
31
注：材料为5A02。
图D.6支架2
注：材料为5A02。
图D.7支架横梁
A.5材料分辨与无机物分辨测试体
材料分辨测试物E用于测试设备的材料分辨能力，测试体由具有不同等效原子序数的尼龙6板、模拟
物板、PVC板、钢板组成。尼龙6板的厚度为25mm，等效原子序数为6.2；模拟物板的厚度为22mm，等效
原子序数为9.8；PVC板的厚度为9.5mm，等效原子序数为14.3；钢板的厚度为1.5mm，等效原子序数为26。GB12664—20XX
32
说明：
①——模拟物板；
②——测试物固定板；
③——PVC板；
④——钢板；
⑤——尼龙板；
⑥——钢阶梯。
图E.1材料分辨与无机物分辨测试体GB12664—20XX
33
注：材料为ABS。
图E.2测试物固定板
注：材料为尼龙6。
图E.3尼龙GB12664—20XX
34
注：材料为模拟物。
图E.4模拟物
注：材料为PVC。
图E.5PVCGB12664—20XX
35
注：材料为SPCC。
图E.6钢板
无机物分辨测试物F，用于测试无机物分辨能力。其由薄钢板阶梯（SPCC）和厚钢板阶梯（45#）
组成，其用盘头螺钉安装在测试体固定板上。
图F.1无机物分辨测试物GB12664—20XX
36A.6有机物与灰度（混合物）分辨体
灰度（混合物）分辨测试物G，用于测试设备灰度和混合物分辨能力，其由测试体固定板、灰度（混
合物）分辨测试物和有机玻璃阶梯组成。
说明：
①——混合物分辨测试体；
②——测试物固定板；
③——有机物分辨测试体。
图G.1有机物与灰度分辨体
注：材料为ABS。
图G.2测试物固定板GB12664—20XX
37灰度（混合物）分辨测试物H，用于测试设备灰度分辨以及混合物分辨的能力。其由薄合金铝阶梯
（5A02）、厚合金铝阶梯（5A02）和铅字组成。厚合金铝阶梯的阶梯厚度为20mm和40mm，测试物用螺
钉安装在测试体固定板上。
图H.1灰度分辨测试
有机物分辨测试物I，由薄有机玻璃阶梯和厚有机玻璃阶梯组成。
图I.1有机物分辨测试物GB12664—20XX
38A.7边框测试物
底部边框测试体J由一块楔形有机玻璃和直径为1mm的钢珠组成，钢珠均匀排列在有机玻璃的沉孔
内，每个沉孔放置30颗钢珠。
图J.1边框测试物
图J.230颗直径为1mm钢珠装配图GB12664—20XX
39
图J.3有机玻璃剖面图
图J.4有机玻璃沉孔放大图GB12664—20XX
40附录B
（规范性）
周围剂量当量率测试方法
方法如下：
a)测量测试环境的天然本底周围剂量当量率，取10个读数的平均值；
b)X射线产生装置放置在空旷的试验场地，距离最近的墙体应在2m以上；
c)放置标准散射体，将标准散射体紧贴射线出束口，使射线沿标准散射体300mm边长方向出束。
通过拼接标准散射体，使散射体的面积不小于射线出束口。
d)将系统设置为应用指标测试中测试5.8.4穿透力指标最大值时的管电压、管电流、脉冲数等；
e)在系统控制区边界、监督区边界和操作人员工作位置至少选定5个测量点。将剂量仪设置为累
积剂量模式，放置在控制区边界、监督区边界、操作人员工作位置等需要测试的位置；
f)将剂量测量设备置于测量点，连续测量本地周围剂量当量，并给出本地周围剂量当量平均值
，取5个读数的平均值。
g)启动便携式安检设备，按设备标准操作方法进行N次典型扫描（即厂家标称的扫描条件，N应
不小于5，扫描结束后，记录测量点累积的周围剂量当量
 及测试时间
——每小时累积周围剂量当量，单位为微希沃特（Sv）；
——N次扫描的累积周围剂量当量，单位为微希沃特（Sv）；
——工作场所本底周围剂量当量率，单位为微希沃特每小时（Sv/h）；
——完成N次扫描的测试时间，单位为小时（h）；
P——每小时内完成的扫描次数，单位为次每小时（次、）。
注：标准散射体材料为软松木，软松木的密度为0.4g/cm3～0.6g/cm3,体积为300mm×300mm×75mm（长×宽×高），
允许公差±5mm。
表B.1测试记录表1—应用指标测试GB12664—20XX
41日期 时间 测试员 设备厂家
型号 序列号 X射线产生装置型号 成像装置型号
X射线产生装置重量 X射线产生装置种类 X射线产生装置与成像装置距离
不可成像的边框宽度
上 下 左 右
有效成像区域 剂量仪型号
泄漏剂量
位置1 位置2 位置3 位置4 位置5
杂散辐射
控制区：
位置1 位置2 位置3 位置4 位置5
监督区：
位置1 位置2 位置3 位置4 位置5
工作位置：
防护装置型号（如有） 尺寸 重量
位置1 位置2 位置3 位置4 位置5
测试项目 测试结果 高压值束流值□脉冲数□所选用的图像处理功能静态□动态□
帧率
穿透力
线分辨力
空间分辨力
穿透分辨力
灰度分辨
表B.2测试记录表2—应用指标测试（适用于多能谱设备）
日期 时间 测试员 设备厂家
型号 序列号 X射线产生装置型号 X射线产生装置厂家
X射线产生装置种类 X射线产生装置与成像装置距离 成像装置种类
成像装置型号
测试项目 最佳测试结果高压值 束流值□脉冲数□ 所选用的图像处理功能
材料分辨
有机物分辨
混合物分辨
无机物分辨
表B.3测试记录表3—典型指标测试GB12664—20XX
42日期 时间 测试员 设备厂家
型号 序列号 X射线产生装置型号 成像装置型号
X射线产生装置重量
测试项目 测试结果 高压值束流值□脉冲数□所选用的图像处理功能静态□动态□
最大帧率
穿透力
线分辨力
空间分辨力
穿透分辨力
灰度分辨力