# Clinical impact of digital and conventional PET control databases for semi-quantitative analysis of brain 18F-FDG digital PET scans

Elise Mairal, Matthieu Doyen, Thérèse Rivasseau-Jonveaux, Catherine Malaplate, Eric Guedj, Antoine Verger

<figure><img src="image_1.png" /></figure>

## To cite this version:

Elise Mairal, Matthieu Doyen, Thérèse Rivasseau-Jonveaux, Catherine Malaplate, Eric Guedj, et al. Clinical impact of digital and conventional PET control databases for semi-quantitative analysis of brain 18F-FDG digital PET scans. EJNMMI Research, 2020, 10 (1), pp.144. 10.1186/s13550-020-00733-y. hal-03222791

HAL Id: hal-03222791

https://amu.hal.science/hal-03222791

Submitted on 10 May 2021

HAL is a multi-disciplinary open access
archive for the deposit and dissemination of sci-
entific research documents, whether they are pub-
lished or not. The documents may come from
teaching and research institutions in France or
abroad, or from public or private research centers.

L'archive ouverte pluridisciplinaire HAL, est
destinée au dépôt et à la diffusion de documents
scientifiques de niveau recherche, publiés ou non,
émanant des établissements d'enseignement et de
recherche français ou étrangers, des laboratoires
publics ou privés.

<figure><img src="image_2.png" /></figure>

Distributed under a Creative Commons Attribution 4.0 International License
ORIGINAL RESEARCH

Open Access

<figure><img src="image_3.png" /></figure>

Check for
updates

# Clinical impact of digital and conventional PET control databases for semi-quantitative analysis of brain ¹⁸F-FDG digital PET scans

Elise Mairal¹, Matthieu Doyen¹,², Thérèse Rivasseau-Jonveaux³, Catherine Malaplate⁴, Eric Guedj⁵
and Antoine Verger¹,²*

**Abstract**

**Purpose:** Digital PET cameras markedly improve sensitivity and spatial resolution of brain ¹⁸F-FDG PET images compared to conventional cameras. Our study aimed to assess whether specific control databases are required to improve the diagnostic performance of these recent advances.

**Methods:** We retrospectively selected two groups of subjects, twenty-seven Alzheimer's Disease (AD) patients and twenty-two healthy control (HC) subjects. All subjects underwent a brain ¹⁸F-FDG PET on a digital camera (Vereos, Philips®). These two group (AD and HC) are compared, using a Semi-Quantitative Analysis (SQA), to two age and sex matched controls acquired with a digital PET/CT (Vereos, Philips®) or a conventional PET/CT (Biograph 6, Siemens®) camera, at group and individual levels. Moreover, individual visual interpretation of SPM T-maps was provided for the positive diagnosis of AD by 3 experienced raters.

**Results:** At group level, SQA using digital controls detected more marked hypometabolic areas in AD (+ 116 cm³ at *p* < 0.001 uncorrected for the voxel, corrected for the cluster) than SQA using conventional controls. At the individual level, the accuracy of SQA for discriminating AD using digital controls was higher than SQA using conventional controls (86% vs. 80%, *p* < 0.01, at *p* < 0.005 uncorrected for the voxel, corrected for the cluster), with higher sensitivity (89% vs. 78%) and similar specificity (82% vs. 82%). These results were confirmed by visual analysis (accuracies of 84% and 82% for digital and conventional controls respectively, *p* = 0.01).

**Conclusion:** There is an urgent need to establish specific digital PET control databases for SQA of brain ¹⁸F-FDG PET images as such databases improve the accuracy of AD diagnosis.

**Introduction**

Digital PET cameras are composed of small digital silicon photomultipliers, which provide digital photon counting with a 1-to-1 crystal coupling, and replace the larger photomultiplier tubes of conventional PET cameras [1, 2]. These digital PET cameras thus provide improvements in detection sensitivity, spatial resolution and signal to noise

ratio and therefore in image quality compared to conventional PET cameras. This has been a particularly significant advance in brain ¹⁸F-FDG PET acquisitions [1, 2].

Brain ¹⁸F-FDG PET is a useful tool for diagnosing neurodegenerative disorders. It is particularly useful in the diagnosis of Alzheimer's Disease (AD) [3], where a visual analysis of brain ¹⁸F-FDG PET images is initially performed to detect the typical AD hypometabolic pattern involving the posterior temporo-parietal association cortex [4, 5]. Semi-quantitative analysis (SQA) has been proposed as an adjunct to this visual analysis since it increases confidence in the diagnostic conclusion

<figure><img src="image_4.png" /></figure>

<footnote>*Correspondence: a.verger@chru-nancy.fr
¹Department of Nuclear Medicine and Nancyclotep Imaging Platform, CHRU Nancy, Université de Lorraine, Rue du Morvan, 54500 Vandoeuvre-les-Nancy, France
Full list of author information is available at the end of the article</footnote>
drawn, particularly at earlier stages of the disease [6]. The importance of SQA has been further underpinned by its inclusion in the recommendations of the European Association of Nuclear Medicine and European Academy of Neurology [3].

The establishment of a well-documented reference database is a crucial step for performing SQA. A comparison to homogeneous control population databases is an important cornerstone to reduce false positives. Indeed, age and sex effects have been reported to influence metabolism distribution in brain <sup>18</sup>F-FDG PET images [7]. In addition, all currently implemented control databases in dedicated software for automated SQA are still based on acquisitions performed with conventional PET [8–13]. It is very likely that improvements in image quality provided by digital PET technology may also influence results of SQA, even though, to the best of our knowledge there is currently no data in the literature which evaluates the potential clinical impact of such an effect.

Our current study aims to assess whether there is a need to establish specific reference control databases which take into account the recent technical advances of SQA on brain <sup>18</sup>F-FDG PET images with respect to the evolution of digital PET.

# Materials and methods

## Subjects

We retrospectively selected two groups of subjects, a group of patients with AD and a group of healthy control subjects (HC). Both groups had undergone a brain <sup>18</sup>F-FDG PET scan on a digital PET/CT (Vereos, Philips®), at the CHRU of Nancy, France, between December 2017 and September 2019.

The AD patients selected fulfilled the NIA-AA 2018 criteria for AD [14]. They exhibited positive cerebrospinal fluid biomarkers with increased phosphorylated Tau protein and reduced beta-amyloid peptide levels measured in the same laboratory (Department of Biochemistry, Molecular Biology and Nutrition, CHRU Nancy, France) using standard cut-offs [15]. AD patients also underwent a routine neurocognitive assessment in the “memory clinic” of the university hospital of Nancy (France).

Healthy control subjects were age and sex matched with patients in the AD group and were also selected retrospectively. All healthy control subjects had undergone a brain <sup>18</sup>F-FDG PET scan for cognitive assessment, but had returned a normal scan by careful visual analysis (EM, AV) and a neuropsychological assessment which was not consistent with a neurodegenerative disorder: (i) normal neuropsychological tests, i.e. MMSE ≥ 27, FAB ≥ 15 and no major depressive disorders and (ii) a

clinical follow-up, of longer than 1 year, which showed a stabilisation and/or improvement of cognitive symptoms.

Our AD and HC groups were further compared to each other and to two control databases (a digital and a conventional database) derived from prospective studies. Individuals from these two control databases had undergone a brain <sup>18</sup>F-FDG PET/CT performed with a conventional camera between October 2009 to May 2012 (n=19, Biograph 6, Siemens®, NCT02858167) or performed with a digital camera between December 2017 to June 2019 (n=20, Vereos, Philips®, NCT03345290) and were age and sex matched with our AD and HC groups. A flowchart summarising the constitution of the different control groups is shown in Fig. 1.

Informed consent was obtained for each participant included in the selected groups. This study was approved on January 16, 2020 by the local ethics committee (NCT04163276, Study ID Numbers: 2019PI238).

## Brain <sup>18</sup>F-FDG PET

The brain <sup>18</sup>F-FDG PET scan was recorded over a 10 (for conventional camera) to 15 min (for digital camera) one bed acquisition, 45–50 min after injection of 4.5 MBq/ kg (conventional camera) or 2–3 MBq/kg (digital camera) of <sup>18</sup>F-FDG. All subjects had fasted at least 6 h prior to receiving the injection and had blood glucose levels < 10 mmol/L. All PET images were reconstructed with iterative OSEM methods, as performed in routine clinical practice, and corrected for scatter, random and attenuation with a CT scan. Reconstructed parameters included 4 iterations and 8 subsets, subsequently displayed in a 168 × 168 matrix with 2.7 × 2.7 × 2.7 voxels for the conventional PET camera [7], and 3 iterations and 15 subsets, subsequently displayed in a 256 × 256 matrix with 1 × 1 × 1 mm³ voxels for the digital PET camera [2].

## Statistical parametric mapping

The <sup>18</sup>F-FDG PET brain images were pre-processed using SPM12 (Wellcome Department of Cognitive Neurology, Institute of Neurology, London, UK) running on Matlab 2018a (MathWorks Inc., Sherborn, MA). After an initial step of approximate manual re-orientation and positioning to MNI space, the spatial normalisation of each PET image into the MNI space was performed by spatial normalisation of the CT scan for each subject provided by the correction of attenuation, using the method and the template of the Clinical Toolbox for SPM (https://www.nitrc.org/projects/clinicaltxb/). The voxel sizes of the written CT images were set to 1 × 1 × 1 mm³ for the digital controls and to 2 × 2 × 2 mm³ for the conventional controls. Each CT spatial normalisation procedure was subsequently applied to the respective PET images. Voxels of PET images recorded with the conventional
<figure><img src="image_5.png" /><figcaption>**Fig. 1** Flowchart for the constitution of the two different groups of Alzheimer's disease and healthy control subjects, and the conventional and digital control databases. Black arrows are representative of analyses performed</figcaption></figure>

camera during this step were therefore resampled from 2.7 × 2.7 × 2.7 mm³ to 2 × 2 × 2 mm³ whereas voxel sizes of the digital databases were not modified (1 × 1 × 1 mm³). To enable voxel-to-voxel analysis with these two control databases, AD and HC group datasets were normalised using the two voxel sizes. Partial volume effect corrections were applied to PET images using the Müller-Gärtner (MG) method provided by the PETPVE12 toolbox [16]. White and grey matter segmentations needed for the 3-compartmental voxel-wise MG method were realised on CT scans using SPM segmentation tools, after careful visual analysis at the individual level to check the accuracy of this segmentation. The cerebellum was used as reference for the intensity normalisation of PET images because normalisations other than the proportional scaling have been proposed [17] and because the cerebellum is associated with a more accurate discrimination of patients with AD compared to controls [18]. All regions of interest (ROIs) used for intensity normalisation (all the cerebellum and vermis ROIs for the cerebellum, and all the pre- and post-central brain areas for the sensorimotor cortex) were extracted from the AAL atlas [19] after spatial normalisation to limit the inter-individual anatomical heterogeneity. Finally, PET images were smoothed with an isotropic 3D Gaussian kernel of 12 mm FWHM to blur individual variations in gyral anatomy. Visual inspections of the images at the different stages of the pre-processing procedure ensured the quality and convergence of the different methods applied.

Semi-Quantitative Analyses (SQA) were performed at the group and individual level on a voxel-by-voxel basis using two-sample t-tests with an inclusive AD mask [20]. At the group level, AD and HC groups were compared with the conventional and digital controls using age and sex as covariates (clusters of decreased metabolic activity observed at *p* < 0.001 for the voxel, cluster volume corrected by using the expected volume provided by SPM and based on the random field theory). We used exclusive masks to compare results obtained with SQA to conventional or digital controls. For AD, an exclusive mask corresponding to the SPM-T map results of SQA to conventional controls was applied to the SQA of digital controls to highlight the additional clusters visualised with the digital system compared to the conventional system (and vice versa for the HC population).

At the individual level, each subject in the AD and HC group, was individually compared to the digital and conventional controls using a fully automated analysis as well as visual analyses (clusters of decreased metabolic activity observed at *p* < 0.005 for the voxel, cluster volumes corrected to 0.8 cm³ [6, 21]). All clusters identified with SPM at the individual level were considered significant for the fully automated analysis.

The precise identification of each structure located by its MNI coordinates, its respective volume, and T-max intensity were extracted by using the report provided by the SPM xjView toolbox (http://www.alivelearn.net/xjvie w).
## Visual ratings at the individual level

For the visual analyses, the SPM T-maps were projected onto three-dimensional rendering of T1-weighted MRI images using SPM surface rendering tool and onto 12 two-dimensional slices of T1-weighted MRI images using the Slice Display toolbox [22] (axial orientation, inter-slice spacing of 1 cm). Representations were reviewed by three experienced observers (EM, EG and AV), who were blinded to the patient's clinical data. Raters were forced to give a dichotomous reading: Alzheimer's disease diagnosis or not pathological. A pattern of diffuse hypometabolic areas within the areas known to be involved in AD (mainly the bilateral posterior associative areas) was considered a positive scan. At the individual level, results were expressed as a consensual analysis for the positive diagnosis of AD.

## Statistical analysis

Categorical variables are expressed as percentages and continuous variables as means and standard deviations. Due to the non-normality of variable distributions, Chi-2 and Kruskal-Wallis tests were performed for comparisons of categorical and continuous variables, respectively. For the comparisons of diagnostic performances at the individual level, Mc Nemar tests were used with corrections for multiple comparisons. A *p* value < 0.05 was considered as significant. All tests were performed with SPSS (SPSS Statistics for Windows, Version 20.0. Armonk, NY: IBM Corp).

## Results

### Population

As detailed in Table 1, no difference in age, sex and educational level were observed between the AD, HC groups, as well as the conventional and digital controls (*p* > 0.27).

As expected, AD patients showed lower levels of MMSE than HC subjects as well as conventional and digital controls (*p* < 0.01).

### At the group level

In patients with AD, SQA using digital controls enabled the detection of more marked hypometabolic areas (+116 cm³) when compared to SQA using conventional controls. There were 3 more clusters of significance identified with SQA using digital controls. The most extensive one involved the bilateral posterior associative areas, including the precuneus and the posterior cingulate (T-voxel max at 7.88). The two others, less extensive and with lower T-voxel max values involved the bilateral anterior associative areas.

Details of significant additional hypometabolic regions obtained with SQA using digital controls in patients with AD are provided in Table 2 and Fig. 2.

Moreover, in healthy control subjects (HC), SQA using conventional controls showed more marked hypometabolic areas (+17 cm³) when compared to SQA using digital controls. There were 3 more significant clusters identified with SQA using conventional controls involving the precuneus, and the bilateral anterior associative areas. Details of significant hypometabolic areas obtained with SQA using conventional controls in healthy control subjects are provided in Table 3 and Fig. 3.

Of note, significant hypermetabolism was also shown with SQA using conventional controls both for comparisons to patients with AD (+21 cm³) and to HC subjects (+115 cm³) whereas no significant hypermetabolism for these comparisons was observed for SQA using digital controls.

For further validation, the AD and HC groups, both acquired with the same digital PET scanner, were

<table><caption>Table 1 Characteristics of the Alzheimer's disease patients, healthy control subject's as well as the conventional and digital control groups</caption><thead><tr><th>Group</th><th>AD (n = 27)</th><th>HC (n = 22)</th><th>Conventional controls (n = 19)</th><th>Digital controls (n = 20)</th><th>p values</th></tr></thead><tbody><tr><td>Age (years old)</td><td>70.9 ± 7.5</td><td>67.7 ± 10.4</td><td>67.2 ± 8.3</td><td>66.6 ± 8.5</td><td>0.27</td></tr><tr><td>Sex (Female)</td><td>14 (52%)</td><td>10 (45%)</td><td>8 (42%)</td><td>8 (40%)</td><td>0.86</td></tr><tr><td>Educational level</td><td></td><td></td><td></td><td></td><td>0.60</td></tr><tr><td>None</td><td>1</td><td>0</td><td>0</td><td>0</td><td></td></tr><tr><td>Primary school</td><td>11</td><td>9</td><td>14</td><td>12</td><td></td></tr><tr><td>High school</td><td>4</td><td>3</td><td>2</td><td>3</td><td></td></tr><tr><td>College</td><td>11</td><td>9</td><td>3</td><td>5</td><td></td></tr><tr><td>NA</td><td>-</td><td>1</td><td>-</td><td>-</td><td></td></tr><tr><td>MMSE</td><td>19.2 ± 5.7*</td><td>28.7 ± 0.9</td><td>29.2 ± 0.8</td><td>29 ± 0.9</td><td>&lt; 0.01*</td></tr></tbody></table>

AD, Alzheimer's disease; HC, healthy control; MMSE, mini-mental state examination score

* *p* < 0.05 for comparisons between groups
<table><caption>**Table 2** Additional AD patient clusters identified with SQA using digital controls as opposed to SQA using conventional controls (anatomical locations, spatial extent of significant clusters in cm³, MNI coordinates, maximal T-scores of the peak voxel) at a T-voxel threshold of 3.3, k cluster size > 1.86 cm³</caption><thead><tr><th>Anatomical location</th><th>Cluster size</th><th>x</th><th>y</th><th>z</th><th>T-score of peak</th></tr></thead><tbody><tr><td>Right and left cerebrum</td><td></td><td></td><td></td><td></td><td></td></tr><tr><td>Cuneus</td><td rowspan="6">105</td><td rowspan="6">-33</td><td rowspan="6">-10</td><td rowspan="6">-44</td><td rowspan="6">7.88</td></tr><tr><td>Precuneus</td></tr><tr><td>Posterior cingulate</td></tr><tr><td>Superior parietal</td></tr><tr><td>Inferior parietal</td></tr><tr><td>Superior temporal</td></tr><tr><td>Middle temporal</td><td rowspan="6">4</td><td rowspan="6">30</td><td rowspan="6">57</td><td rowspan="6">-2</td><td rowspan="6">5.29</td></tr><tr><td>Inferior temporal</td></tr><tr><td>Superior occipital</td></tr><tr><td>Middle occipital</td></tr><tr><td>Inferior occipital</td></tr><tr><td>Right cerebrum</td></tr><tr><td>Superior frontal</td><td rowspan="3">7</td><td rowspan="3">-36</td><td rowspan="3">55</td><td rowspan="3">0</td><td rowspan="3">4.16</td></tr><tr><td>Middle frontal</td></tr><tr><td>Inferior frontal</td></tr><tr><td>Left cerebrum</td><td></td><td></td><td></td><td></td><td></td></tr><tr><td>Superior frontal</td><td rowspan="3">7</td><td rowspan="3">-36</td><td rowspan="3">55</td><td rowspan="3">0</td><td rowspan="3">4.16</td></tr><tr><td>Middle frontal</td></tr><tr><td>Inferior frontal</td></tr></tbody></table>

compared at the group-level. The volume of cluster significance was similar to that observed in SQA using digital controls for patients with AD (195 cm³ with a T-voxel max value of significance at 7.48, vs. 185 cm³ with a T-voxel max value of significance at 7.88 for the comparison between patients with AD and the digital controls).

### At the individual level

The accuracy of SQA for discriminating AD using digital controls was higher than with conventional controls (86 vs. 80%), with a higher sensitivity (89 vs. 78%) and similar specificity (82% vs. 82%) achieved with the fully automated analysis.

These results were confirmed by the visual analysis with an accuracy, sensitivity and specificity of respectively 84%, 85%, 82% for SQA using digital and 82%, 67%, 100% for SQA using conventional controls.

Detailed diagnostic performances for both SQA using conventional and digital controls are reported in Table 4.

An illustration of SPM-T map images used for the visual analysis is displayed in Fig. 4. Figure 5 provides a galleria of individual patients with both SQA to conventional and digital controls.

<figure><img src="image_6.png" /><figcaption>**Fig. 2** Anatomical localisation of areas of decreased metabolic activity in AD patients (*p* < 0.001, uncorrected, *k* > 1.86 cm³) for SQA using digital controls (left panel), SQA using conventional controls (middle panel), and the significant additional hypometabolic areas obtained with SQA using digital controls compared to SQA using conventional controls (right panel), projected onto 3D volume rendering, spatially normalised and smoothed into the standard SPM template (T-voxel threshold value of 3.3)</figcaption></figure>
<table><caption>**Table 3** Additional HC patient clusters identified with SQA using conventional controls as opposed to SQA using digital controls (anatomical locations, spatial extent of significant clusters in cm³, MNI coordinates, maximal T-scores of the peak voxel) at a T-voxel threshold of 3.3, k cluster size > 1.45 cm³</caption><thead><tr><th>Anatomical location</th><th>Cluster size</th><th>x</th><th>y</th><th>z</th><th>T-score of peak</th></tr></thead><tbody><tr><td>Right cerebrum</td><td></td><td></td><td></td><td></td><td></td></tr><tr><td>Middle frontal</td><td>5</td><td>34</td><td>2</td><td>38</td><td>5.73</td></tr><tr><td>Left cerebrum</td><td></td><td></td><td></td><td></td><td></td></tr><tr><td>Middle frontal</td><td>7</td><td>-24</td><td>2</td><td>46</td><td>5.01</td></tr><tr><td>Right cerebrum</td><td></td><td></td><td></td><td></td><td></td></tr><tr><td>Precuneus</td><td>5</td><td>10</td><td>-52</td><td>22</td><td>5.96</td></tr></tbody></table>

Discussion

This study aimed to assess whether the event of SQA and the evolution of digital PET imaging requires the control databases to evolve concomitantly. Diagnostic performances observed at the group and individual level show that the diagnostic accuracy of SQA on digital controls is improved compared to SQA on conventional controls, particularly as it relates to the detection sensitivity

of AD. This observation is an argument which supports the development of digital control databases for SQA of brain ¹⁸F-FDG PET images for clinical practice.

Digital PET technology is associated with improvements in image quality, specifically better spatial resolution and signal-to-noise ratios compared to conventional PET cameras [2]. These distinct image qualities lead to problematic head-to-head comparisons between digital and conventional PET images as reflected in our study by the relatively poor diagnostic performance obtained for SQA using conventional controls for discriminating AD. However, all currently implemented control databases in dedicated software for automated SQA in clinical practice still rely on conventional PET control images [9, 13]. Of course, it is now possible to implement local databases in the majority of these types of software, but establishing control databases acquired with digital PET technology remains an extensive undertaking, particularly because it involves a relative recently implemented technology.

Our present study shows that implementing a control database acquired with digital PET technology yields an increase in the detection sensitivity of AD patients, not only at the group level (+116 cm³ of detected hypometabolism volume, Table 2) but also at the individual

<figure><img src="image_7.png" /><figcaption>**Fig. 3** Anatomical localisation of areas of decreased metabolic activity in HC subjects (*p* < 0.001, uncorrected, *k* > 1.45 cm³), with SQA using digital controls (left panel) and SQA using conventional controls (right panel), projected onto 3D volume rendering, spatially normalised and smoothed into the standard SPM template (T-voxel threshold value of 3.3)</figcaption></figure>
<table><caption>**Table 4** Diagnostic performances of SQA using conventional and digital controls at the individual level with fully automated and visual analyses</caption><thead><tr><th></th><th>Accuracy (%)</th><th>Sensitivity (%)</th><th>Specificity (%)</th><th>p value</th></tr></thead><tbody><tr><td>Fully automated</td><td></td><td></td><td></td><td>&lt; 0.01*</td></tr><tr><td>SQA using conventional controls</td><td>80</td><td>78</td><td>82</td><td></td></tr><tr><td>SQA using digital controls</td><td>86</td><td>89</td><td>82</td><td></td></tr><tr><td>Visual</td><td></td><td></td><td></td><td>= 0.01*</td></tr><tr><td>SQA using conventional controls</td><td>82</td><td>67</td><td>100</td><td></td></tr><tr><td>SQA using digital controls</td><td>84</td><td>85</td><td>82</td><td></td></tr></tbody></table>

<figure><img src="image_8.png" /><figcaption>**Fig. 4** Example of SPM-T map images used for the visual analysis (semi-quantitative analysis to digital controls) of a patient with AD (79-year-old man with MMSE score of 10). SPM-T maps are projected onto two-dimensional slices of T1-weighted MRI (from the base to the top of the skull, left panel) and 3D-rendered volumes (right panel)</figcaption></figure>

level (sensitivity of detection increased from 78 to 89%, Table 4). A high detection sensitivity is primordial in the diagnosis of AD since ¹⁸F-FDG PET is a biomarker of neurodegeneration, which contributes to the ATN classification, the N biomarker being directly associated with cognitive impairment in patients suspected to have neurodegenerative diseases [14].

All results in the current study were initially obtained using a fully automated analysis, which supports the objective nature of our observations in both the group and individual level analyses. This original fully automated methodology, which necessitated an adaptation of the levels of significance to detect anomalies, was exclusively based on the SPM software. From a clinical

<footnote>* For the comparison between SQA using conventional and digital controls</footnote>
<figure><img src="image_9.png" /></figure>

standpoint and at the individual level, this fully auto-
mated analysis was nevertheless consistent with the
visual analysis, using a methodology that is very simi-
lar to that applied in previously published SQA stud-
ies [6, 12]. By using this visual analysis, the diagnostic

**Fig. 5** Galleria of individual examples with both semi-quantitative analyses to conventional and digital controls. Three examples of SPM-T map images from semi-quantitative analyses to conventional (left panel) and digital (right panel) controls of patients with AD (a 74-year-old woman with MMSE score of 19, b 70-year-old woman with MMSE score of 21, c 66-year-old man with MMSE score of 23). One example of SPM-T map images from semi-quantitative analyses to conventional (left panel) and digital (right panel) controls from a healthy subject (d 54-year-old woman with MMSE score of 29). SPM-T maps are projected onto two-dimensional slices of T1-weighted MRIs (from the base to the top of the skull)

performance of SQA for discriminating AD with the
digital controls observed in our study (accuracy, sen-
sitivity and specificity of respectively 84, 85 and 82%)
was within the range of previously reported results
(70–97.5% for accuracy, 62.3–96% for sensitivity and
84–99% for specificity) [6, 8–12, 23–25].

SQA at the individual level was also performed using
an intensity normalisation based on the sensorimotor
cortex, which has been suggested to improve diagnostic
performance of SQA [26]. The finds results were com-
parable to intensity normalisation on cerebellum. SQA
on digital controls show respectively accuracy, sensitiv-
ity and specificity 84%, 89% and 77% (vs. 86%, 89% and
82%). SQA on conventional controls show respectively
accuracy, sensitivity and specificity 73%, 52% and 100%
(vs. 80% 78% and 82%).

The post-filter used (Gaussian kernel of 12 mm
FWHM) can be decreased relative to the size of the
voxels used particularly for digital PET scans. How-
ever, when smoothing images with a Gaussian kernel of
4 mm FWHM, the accuracy to detect AD, respectively
versus digital and conventional controls, remained
unchanged (86% vs. 86% and 82% vs. 80%).

The main limitation of our study results from the fact
that controls included in the conventional and digital
control databases were different individuals. From an
ethical perspective, it remains problematic to estab-
lish control databases acquired in parallel with both
the conventional and digital PET systems, even if it
would be feasible to scan the same set of HC subjects
within a few months on two different scanners using
the half-dose permitted by the high sensitivity of the
digital PET. It should however be noted that controls
included in our conventional and digital databases did
not exhibit any differences in age, sex, MMSE and edu-
cational level when compared to each other, or when
compared to the AD and HC groups. In addition, these
two, distinct conventional and digital control databases
are representative samples from current daily clinical
practice. The main objective of the current study was to
assess whether there is indeed a requirement to estab-
lish digital control databases when acquisitions are
performed with the new digital PET system. A second-
ary issue that may be addressed is that the sample size
of conventional and digital control databases is rather
small (n = 19 and 20). This number of controls is nev-
ertheless known to be sufficient to accurately perform
group analyses with SPM [27].

Overall, in light of recent digital PET technology devel-
opments and considering that SQA is now clearly recom-
mended for brain ¹⁸F-FDG PET image analysis, there is an
urgent need to establish digital PET control databases for
SQA of brain ¹⁸F-FDG PET images. This would be particu-
larly helpful for improving the sensitivity required to detect
AD patients. Large healthy control databases should be
constituted and shared through the multicentre commu-
nity using standardised imaging protocols.

Acknowledgements

The authors thank the Nancyclotep Imaging Platform for their support.

Authors' contributions

All authors contributed significantly to the analysis and interpretation of the data (EM, MD, TJ, CM), to the writing of the manuscript (EM, MD, AV) and to the revision of the manuscript (TJ, CM, EG, AV). All authors read and approved the final manuscript.

Funding

None.

Availability of data and materials

The data that support the findings of this study are available on request from
the corresponding author (AV).

Code availability

Not applicable.

Ethics approval and consent to participate

All procedures performed in this study involving human participants were
in accordance with the ethical standards of the institutional and/or national
research committees and with the 1964 Helsinki declaration and its latest
amendments or comparable ethics standards. Subjects included in the two
control databases were derived from prospective studies (NCT02858167 and
NCT03345290).

Consent for publication

Not applicable.

Competing interests

The authors disclose no potential conflicts of interest related to the present
work.

Informed consent

Informed consent was obtained for each subject included in the two retro-
spectively selected groups. This study was approved on January 16, 2020 by
the local ethics committee (NCT04163276, Study ID Numbers: 2019PI238).

Author details

Received: 31 July 2020 Accepted: 23 November 2020

Published online: 30 November 2020

References

1. Salvadori J, Odille F, Verger A, Olivier P, Karcher G, Marie P-Y, et al. Head-to-head comparison between digital and analog PET of human and phantom images when optimized for maximizing the signal-to-noise ratio from small lesions. EJNMMI Phys. 2020;7(1):11.

2. Salvadori J, Imbert L, Perrin M, Karcher G, Lamiral Z, Marie P-Y, et al. Head-to-head comparison of image quality between brain 18F-FDG images recorded with a fully digital versus a last-generation analog PET camera. EJNMMI Res. 2019;9(1):61.

3. Nobili F, Arbizu J, Bouwman F, Drzezga A, Agosta F, Nestor P, et al. European Association of Nuclear Medicine and European Academy of Neurology recommendations for the use of brain 18 F-fluorodeoxyglucose positron emission tomography in neurodegenerative cognitive impairment and dementia: Delphi consensus. Eur J Neurol. 2018;25(10):1201–17.

4. Nestor PJ, Altomare D, Festari C, Altomare D, Walker Z, Bouwman F, Rivolta J, et al. Clinical utility of FDG-PET for the clinical diagnosis in MCI. Eur J Nucl Med Mol Imaging. 2018;45(9):1497–508.

5. Arbizu J, Festari C, Altomare D, Walker Z, Bouwman F, Rivolta J, et al. Clinical utility of FDG-PET for the differential diagnosis among the main forms of dementia. Eur J Nucl Med Mol Imaging. 2018;45(9):1509–25.

6. Perani D, Della Rosa PA, Cerami C, Gallivanone F, Fallanca F, Vanoli EG, et al. Validation of an optimized SPM procedure for FDG-PET in dementia diagnosis in a clinical setting. NeuroImage Clin. 2014;6:445–54.

7. Van Der Gucht A, Verger A, Guedj E, Malandain G, Hossu G, Yagdigul Y, et al. Age-related changes in FDG brain uptake are more accurately assessed when applying an adaptive template to the SPM method of voxel-based quantitative analysis. Ann Nucl Med. 2015;29(10):921–8.

8. Nobili F, Festari C, Altomare D, Agosta F, Orini S, Van Laere K, et al. Automated assessment of FDG-PET for differential diagnosis in patients with neurodegenerative disorders. Eur J Nucl Med Mol Imaging. 2018;45(9):1557–66.

9. Ishii K, Kono AK, Sasaki H, Miyamoto N, Fukuda T, Sakamoto S, et al. Fully automatic diagnostic system for early- and late-onset mild Alzheimer's disease using FDG PET and 3D-SSP. Eur J Nucl Med Mol Imaging. 2006;33(5):575–83.

10. Burdette JH, Minoshima S, Vander Borght T, Tran DD, Kuhl DE. Alzheimer disease: improved visual interpretation of PET images by using three-dimensional stereotaxic surface projections. Radiology. 1996;198(3):837–43.

11. Kono AK, Ishii K, Sofue K, Miyamoto N, Sakamoto S, Mori E. Fully automatic differential diagnosis system for dementia with Lewy bodies and Alzheimer's disease using FDG-PET and 3D-SSP. Eur J Nucl Med Mol Imaging. 2007;34(9):1490–7.

12. Lehman VT, Carter RE, Claassen DO, Murphy RC, Lowe V, Petersen RC, et al. Visual assessment versus quantitative three-dimensional stereotaxic surface projection fluorodeoxyglucose positron emission tomography for detection of mild cognitive impairment and Alzheimer disease. Clin Nucl Med. 2012;37(8):721–6.

13. Sun X, Liu F, Liu Q, Gai Y, Ruan W, Wimalarathne DN, et al. Quantitative research of 11C-CFT and 18F-FDG PET in Parkinson's disease: a pilot study with NeuroQ software. Front Neurosci. 2019;13:299.

14. Jack CR, Bennett DA, Blennow K, Carrillo MC, Dunn B, Haeberlein SB, et al. NIA-AA research framework: toward a biological definition of Alzheimer's disease. Alzheimers Dement J Alzheimers Assoc. 2018;14(4):535–62.

15. Manca C, Rivasseau Jonveaux T, Roch V, Marie P-Y, Karcher G, Lamiral Z, et al. Amyloid PETs are commonly negative in suspected Alzheimer's disease with an increase in CSF phosphorylated-tau protein concentration but an Aβ42 concentration in the very high range: a prospective study. J Neurol. 2019;266(7):1685–92.

16. Gonzalez-Escamilla G, Lange C, Teipel S, Buchert R, Grothe MJ, Alzheimer's Disease Neuroimaging Initiative. PETPVE12: an SPM toolbox for partial volume effects correction in brain PET—application to amyloid imaging with AV45-PET. NeuroImage. 2017;147:669–77.

<footnote n="1">¹ Department of Nuclear Medicine and Nancyclotep Imaging Platform, CHRU Nancy, Université de Lorraine, Rue du Morvan, 54500 Vandoeuvre-les-Nancy, France. ² IADI, INSERM U1254, Université de Lorraine, 54000 Nancy, France.</footnote>

<footnote n="3">³ Clinical Memory and Research Center, Department of Geriatrics, CHRU Nancy, Université de Lorraine, 2LPN EA 7489, 54000 Nancy, France. ⁴ Department of Biochemistry, Molecular Biology and Nutrition CHRU Nancy, Université de Lorraine, 54000 Nancy, France. ⁵ APHM, CNRS, Centrale Marseille, Institut Fresnel, Timone Hospital, CERIMED, Nuclear Medicine Department, Aix Marseille Univ, Marseille, France.</footnote>
17. Minoshima S, Frey KA, Foster NL, Kuhl DE. Preserved pontine glucose metabolism in Alzheimer disease: a reference region for functional brain image (PET) analysis. J Comput Assist Tomogr. 1995;19(4):541-7.

18. Dukart J, Mueller K, Horstmann A, Vogt B, Frisch S, Barthel H, et al. Differential effects of global and cerebellar normalization on detection and differentiation of dementia in FDG-PET studies. Neurolmage. 2010;49(2):1490-5.

19. Tzourio-Mazoyer N, Landeau B, Papathanassiou D, Crivello F, Etard O, Delcroix N, et al. Automated anatomical labeling of activations in SPM using a macroscopic anatomical parcellation of the MNI MRI single-subject brain. Neurolmage. 2002;15(1):273-89.

20. Herholz K, Salmon E, Perani D, Baron JC, Holthoff V, Frölich L, et al. Discrimination between Alzheimer dementia and controls by automated analysis of multicenter FDG PET. Neurolmage. 2002;17(1):302-16.

21. Della Rosa PA, Cerami C, Gallivanone F, Prestia A, Caroli A, Castiglioni I, et al. A standardized [18F]-FDG-PET template for spatial normalization in statistical parametric mapping of dementia. Neuroinformatics. 2014;12(4):575-93.

22. Zandbelt B. bramzandbelt/slice_display [Internet]. 2019 [cité 13 mai 2020]. Disponible sur: https://github.com/bramzandbelt/slice_display

23. Morbelli S, Brugnolo A, Bossert I, Buschiazzo A, Frisoni GB, Galluzzi S, et al. Visual versus semi-quantitative analysis of 18F-FDG-PET in amnestic MCI: an European Alzheimer's Disease Consortium (EADC) project. J Alzheimers Dis. 2015;44(3):815-26.

24. Rabinovici GD, Rosen HJ, Alkalay A, Kornak J, Furst AJ, Agarwal N, et al. Amyloid vs FDG-PET in the differential diagnosis of AD and FTLD. Neurology. 2011;77(23):2034-42.

25. Ng S, Villemagne VL, Berlangieri S, Lee S-T, Cherk M, Gong SJ, et al. Visual assessment versus quantitative assessment of 11C-PIB PET and 18F-FDG PET for detection of Alzheimer's disease. J Nucl Med Off Publ Soc Nucl Med. 2007;48(4):547-52.

26. Yakushev I, Landvogt C, Buchholz H-G, Fellgiebel A, Hammers A, Scheu-rich A, et al. Choice of reference area in studies of Alzheimer's disease using positron emission tomography with fluorodeoxyglucose-F18. Psychiatry Res. 2008;164(2):143-53.

27. Signorini M, Paulesu E, Friston K, Perani D, Colleluori A, Lucignani G, et al. Rapid assessment of regional cerebral metabolic abnormalities in single subjects with quantitative and nonquantitative [18F]FDG PET: a clinical validation of statistical parametric mapping. Neurolmage. 1999;9(1):63-80.

## Publisher's Note

Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.

Submit your manuscript to a SpringerOpen journal and benefit from:

* Convenient online submission

* Rigorous peer review

* Open access: articles freely available online

* High visibility within the field

* Retaining the copyright to your article

Submit your next manuscript at ► springeropen.com