# Photon-counting versus energy-integrating CT of abdomen-pelvis: a phantom study on the potential for reducing iodine contrast media

Djamel Dabli, Maxime Pastor, Sebastian Faby, Julien Erath, Cédric Croisille, F. Pereira, Jean-Paul Beregi, Joël Greffier

► To cite this version:

Djamel Dabli, Maxime Pastor, Sebastian Faby, Julien Erath, Cédric Croisille, et al.. Photon-counting versus energy-integrating CT of abdomen-pelvis: a phantom study on the potential for reducing iodine contrast media. European Radiology Experimental, 2025, 9, pp.36. 10.1186/s41747-025-00573-2. hal-05066215

HAL Id: hal-05066215

https://hal.science/hal-05066215v1

Submitted on 13 May 2025

HAL is a multi-disciplinary open access
archive for the deposit and dissemination of sci-
entific research documents, whether they are pub-
lished or not. The documents may come from
teaching and research institutions in France or
abroad, or from public or private research centers.

L'archive ouverte pluridisciplinaire HAL, est
destinée au dépôt et à la diffusion de documents
scientifiques de niveau recherche, publiés ou non,
émanant des établissements d'enseignement et de
recherche français ou étrangers, des laboratoires
publics ou privés.

<figure><img src="image_1.png" /></figure>

Distributed under a Creative Commons Attribution 4.0 International License
# Photon-counting *versus* energy-integrating CT of abdomen-pelvis: a phantom study on the potential for reducing iodine contrast media

<figure><img src="image_2.png" /></figure>

Check for
updates

Djamel Dabli¹* , Maxime Pastor¹, Sebastian Faby², Julien Erath², Cédric Croisille², Fabricio Pereira³,
Jean-Paul Beregi¹ and Joël Greffier¹

**Abstract**

**Background** To assess the potential of virtual monoenergetic images (VMIs) on a photon-counting computed tomography (PCCT) for reducing the amount of injected iodine contrast media compared to an energy-integrating CT (EICT).

**Methods** A multienergy phantom was scanned with a PCCT and EICT at 11 mGy with abdomen-pelvis examination parameters. VMIs were generated at 40 keV, 50 keV, 60 keV, and 70 keV. For all VMIs, the contrast-to-noise ratio (CNR) of iodine inserts with concentrations of 1 mg/mL, 2 mg/mL, 5 mg/mL, 10 mg/mL, and 15 mg/mL was calculated by dividing the signal difference between HU in iodine inserts versus solid water by the noise value assessed on solid water. The potential reduction in iodine media was calculated by the rate of reduction in iodine concentration with PCCT while maintaining the same CNR obtained with EICT for the reference concentration.

**Results** Significantly higher CNR values were found with PCCT at all VMI energy levels for iodine concentrations above 1 mg/mL. The highest reduction was observed at 40 keV, with a value of 48.9 ± 1.6% (mean ± standard deviation). It decreased as the energy level increased, by 38.5 ± 0.5%, and 30.8 ± 0.8% for 50 and 60 keV, respectively. For 70 keV, the potential reduction of 24.4 ± 1.1% was found for iodine concentrations above 1 mg/mL. This reduction reached 57 ± 2.3% at 40 keV with PCCT compared to 60 keV with EICT.

**Conclusion** For abdomen-pelvis protocols, the use of VMIs with PCCT significantly improved the CNR of iodine, offering the potential to reduce the required contrast medium.

**Relevance statement** The use of VMIs with PCCT may reduce the quantity of iodine contrast medium to be injected compared with EICT, limiting costs, the risk of adverse effects, and the amount of contrast agent released into the wastewater.

**Key Points**

* PCCT improves the image quality of VMIs.

* PCCT offers the potential for reducing the amount of injected contrast medium.

* PCCT potential for reducing the injected contrast medium depends on energy level.

**Keywords** Abdomen, Contrast media, Iodine compounds, Pelvis, Tomography (x-ray computed)

<figure><img src="image_3.png" /></figure>

<footnote>*Correspondence:
Djamel Dabli
djamel.dabli@chu-nimes.fr
Full list of author information is available at the end of the article</footnote>
**Graphical Abstract**

**Photon-counting versus energy-integrating CT
of abdomen-pelvis: a phantom study on the
potential for reducing iodine contrast media**

<figure><img src="image_4.png" /></figure>

EUROPEAN SOCIETY
OF RADIOLOGY

* Photon-counting CT (PCCT) improves the image quality of virtual monoenergetic images.

* PCCT offers a significant potential for reducing the amount of contrast medium.

* PCCT potential for reducing the iodine media decreased as the energy level increased.

<figure><img src="image_5.png" /></figure>

CNR variation on VMIs according to iodine concentrations for PCCT and EICT: a. 40 keV, b. 50 keV, c. 60 keV, d. 70 keV

**A potential for reducing the amount of iodine contrast media
was found with PCCT at reference radiation dose**

<figure><img src="image_6.png" /></figure>

European
Radiology
EXPERIMENTAL

Eur Radiol Exp (2025) Dabli D, Pastor M, Faby S et al;
DOI: 10.1186/s41747-025-00573-2

**Background**

Computed tomography (CT) examinations are often
combined with an injection of iodinated contrast media
(ICM) to enhance the contrast of structures or lesions [1].
However, the risk of adverse effects from ICM must be
taken into account in addition to the risk associated with
x-ray exposure, especially in patients suffering from renal
insufficiency [2–4].

The latest technological developments in CT acquisi-
tion and reconstruction methods [5–8] have made it
possible to reduce both the radiation dose and the amount
of ICM. Among these developments, dual-energy CT
(DECT) equipped with energy-integrated detectors plays
an important role. DECT utilizes two distinct spectra,
characterized by lower and higher average energies
compared to conventional CT, allowing differentiation of
materials based on attenuation [9–13]. It is then possible
to generate several types of images, notably, virtual
monoenergetic images (VMIs). These images are visua-
lized similarly to those that would have been obtained
with a monochromatic beam which enhances the contrast
of iodine in the tissues using low energy levels [14],
improves the detection and characterization of abdominal
lesions [14–18], and offers significant potential for redu-
cing the amount of contrast media [19–21]. However, a

phantom study by Van Hamersvelt et al [21] showed that
iodine concentration could be reduced by up to 60% using
VMIs at 40 keV on a dual-layer CT in angiography CT
imaging of the aorta and coronary arteries.

Since 2021, a new generation of CT systems equipped
with photon-counting detectors has become available.
These new detectors are able to count photons according
to their energy and have improved noise performance by
suppressing electronic noise. They also have a better
spatial resolution than energy-integrating detectors,
thanks to their smaller detector elements [22–25]. Several
studies have demonstrated a significant improvement in
the image quality of VMIs with photon-counting CT
(PCCT) and the potential of radiation dose reduction
[26–31] compared to energy-integrating detector CT
(EICT). Some studies have also demonstrated the poten-
tial of PCCT for reducing the amount of ICM estimated at
42% for pulmonary angiography CT [32], 40–50% for
coronary CT angiography [33–35], and 25% for thor-
acoabdominal aorta CT angiography [36, 37]. However,
these studies focused on chest and cardiac examinations.
To the best of our knowledge, so far there have been no
studies exploring the potential of VMIs generated with
PCCT for reducing the amount of ICM injection in
abdomen-pelvis examinations.
<figure><img src="image_7.png" /><figcaption>**Fig. 1** VMI at 50 keV with a region of interest placed in iodine and solid water inserts</figcaption></figure>

The reduction in the amount of ICM used needs to be consistent with maintaining sufficient image quality for diagnostic purposes. This means maintaining a sufficient contrast-to-noise ratio (CNR) for structures containing iodine. Several phantom studies have shown a linear relationship between CNR and the iodine content of structures for fixed mean beam energy [38, 39]. As a result, CNR can be used as a metric to assess the amount of contrast medium reduction possible while maintaining image quality similar to a reference image.

The purpose of this phantom study was to evaluate the potential of PCCT to reduce the amount of injected ICM compared to EICT, using DECT and VMI for abdomen-pelvis protocols, based on the CNR metric.

# Methods

## Phantom

Multienergy CT phantom commercialized by Sun Nuclear (Melbourne, Australia) of 20-cm diameter, with an elliptical ring (30 × 40 × 15 cm³) was scanned. The phantom is composed of a 20-cm diameter cylinder inserted on an elliptical ring allowing the phantom dimensions to be extended to represent a patient's abdomen by inserting the 20-cm cylinder into the insert. The whole phantom with the elliptical insert measures 40 cm wide, 30 cm high, and 15 cm depth. The background material of the phantom is solid water. It contains 27 holes for solid inserts of 28 mm in diameter simulating different materials (Fig. 1).

Five iodine inserts with concentrations ranging from 1 mg/mL to 15 mg/mL and a solid water insert were placed in the center part of the phantom.

## Image acquisition and reconstruction

Acquisitions were performed on a PCCT (NAEOTOM Alpha, Siemens Healthineers, Forchheim, Germany)

<table><caption>**Table 1** Acquisition and reconstruction parameters used on energy-integrated and PCCT</caption><thead><tr><th>Scanner model</th><th>SOMATOM Force</th><th>NAEOTOM Alpha</th></tr></thead><tbody><tr><td>Detector</td><td>Stellar (EID)</td><td>QuantaMax (PCD)</td></tr><tr><td>Collimation (mm)</td><td>128 × 0.6 mm</td><td>144 × 0.4 mm</td></tr><tr><td>Pitch</td><td>0.8</td><td>0.8</td></tr><tr><td>Rotation time (s)</td><td>0.5</td><td>0.5</td></tr><tr><td>Tube voltage (kV)</td><td>100/Sn150</td><td>140</td></tr><tr><td>Displayed CTDI<sub>vol</sub> (mGy)</td><td>10.96</td><td>11.00</td></tr><tr><td>Tube current (mAs) (tube A/tube B for EID and tube A for PCD)</td><td>189/95</td><td>96</td></tr><tr><td>Reconstruction kernel</td><td>Qr36</td><td>Qr36</td></tr><tr><td>Reconstruction algorithm</td><td>ADMIRE</td><td>QIR</td></tr><tr><td>Level</td><td>3</td><td>3</td></tr></tbody></table>

ADMIRE Advanced modeled iterative reconstruction, CTDI<sub>vol</sub> Volume CT dose index, EID Energy-integrated, PCD Photon-counting, QIR Quantum iterative reconstruction

equipped with the photon-counting detector and on an EICT corresponding to the third-generation dual-source CT (SOMATOM Force, Siemens Healthineers). Two tube-detector pairs were available for PCCT and EICT. On the EICT, spectral acquisition is carried out using the two x-ray tubes with two different voltages, but on the PCCT this acquisition is carried out using a single x-ray tube [9].

Then, the default acquisition parameters for classical abdomen-pelvis DECT examinations were used on each scanner (Table 1). For EICT, 100/Sn150 kVp (tube A/tube B) was used, and 140 kVp for PCCT. For both scanners, the tube current value (mAs) was set to obtain a volume CT dose index of around 11 mGy, corresponding to the national guide value for abdomen-pelvis examinations. Five acquisitions were repeated for each CT scan.

Raw data were reconstructed for both CT scans using the soft tissue quantification kernel (Qr36), a slice thickness of 1 mm (in 1-mm increments), and a 420-mm field of view. Level 3 of the Advanced Modeled Iterative Reconstruction algorithm was used for EICT and level 3 of Quantum Iterative Reconstruction algorithm for PCCT.

For all CT acquisitions, VMIs were generated at four low energy levels (40 keV, 50 keV, 60 keV, and 70 keV) using the “Monoenergetic+” application of the *syngo.via* postprocessing software (version VB60A_HF03, Siemens Healthineers).

## CNR measurement

Several studies have reported that ICM concentration values in abdominal structures ranged from 5 mg/mL to 10 mg/mL in the aorta [40–42] and 1.23–4.31 mg/mL in
<table><caption>**Table 2** Contrast values between iodine inserts and solid water insert according to monoenergetic levels for EICT and PCCT</caption><thead><tr><th>Inserts</th><th>Energy level (keV)</th><th>EICT</th><th>PCCT</th><th>p-value</th></tr></thead><tbody><tr><td rowspan="4">Iodine at 15 mg/mL</td><td>40</td><td>1234.2 ± 8.2</td><td>1242.0 ± 5.9</td><td>0.091</td></tr><tr><td>50</td><td>807.4 ± 5.4</td><td>813.1 ± 3.5</td><td>0.069</td></tr><tr><td>60</td><td>548.3 ± 3.9</td><td>552.1 ± 1.8</td><td>0.056</td></tr><tr><td>70</td><td>388.5 ± 3.1</td><td>389.0 ± 4.6</td><td>0.855</td></tr><tr><td rowspan="4">Iodine at 10 mg/mL</td><td>40</td><td>820.7 ± 9.4</td><td>839.5 ± 5.9</td><td>0.057</td></tr><tr><td>50</td><td>538.5 ± 6.1</td><td>549.5 ± 3.2</td><td>0.061</td></tr><tr><td>60</td><td>367.2 ± 4.1</td><td>372.9 ± 1.4</td><td>0.052</td></tr><tr><td>70</td><td>261.6 ± 3.0</td><td>262.0 ± 4.5</td><td>0.892</td></tr><tr><td rowspan="4">Iodine at 5 mg/mL</td><td>40</td><td>416.3 ± 8.9</td><td>422.9 ± 4.5</td><td>0.206</td></tr><tr><td>50</td><td>273.3 ± 6.1</td><td>277.6 ± 2.3</td><td>0.237</td></tr><tr><td>60</td><td>186.7 ± 4.5</td><td>189.3 ± 1.0</td><td>0.274</td></tr><tr><td>70</td><td>133.3 ± 3.5</td><td>133.0 ± 4.6</td><td>0.906</td></tr><tr><td rowspan="4">Iodine at 2 mg/mL</td><td>40</td><td>172.0 ± 11.7</td><td>147.5 ± 4.7</td><td>0.052</td></tr><tr><td>50</td><td>111.8 ± 7.1</td><td>98.9 ± 3.0</td><td>0.056</td></tr><tr><td>60</td><td>75.7 ± 3.8</td><td>69.0 ± 1.9</td><td>0.063</td></tr><tr><td>70</td><td>53.4 ± 2.3</td><td>48.3 ± 5.5</td><td>0.136</td></tr><tr><td rowspan="4">Iodine at 1 mg/mL</td><td>40</td><td>87.9 ± 11.8</td><td>54.5 ± 5.8</td><td><strong>0.009</strong></td></tr><tr><td>50</td><td>59.0 ± 8.0</td><td>36.9 ± 2.9</td><td><strong>0.007</strong></td></tr><tr><td>60</td><td>42.2 ± 5.0</td><td>26.1 ± 1.3</td><td><strong>0.001</strong></td></tr><tr><td>70</td><td>31.9 ± 3.2</td><td>17.6 ± 5.0</td><td><strong>0.004</strong></td></tr></tbody></table>

Significant *p*-values are shown in bold

other abdominal organs or lesions [43–46]. Thus, iodine inserts with concentrations of 1 mg/mL, 2 mg/mL, 5 mg/mL, 10 mg/mL, and 15 mg/mL were used in this study to be representative of clinical concentrations. The solid water insert was also used as a reference representing tissues not containing iodine. For each energy level, six circular regions of interest (ROIs) of 2 cm² were placed in the center of the solid water insert and on five iodine inserts on the central slice of the phantom (Fig. 1). For each ROI, the mean HU value was measured on each VMI using the “Monoenergetic +” application in the syngo.via software.

The CNR was defined as the signal difference between HU in iodine inserts versus background (solid water), divided by background noise. For each iodine insert, the iodine CNR was calculated using the following Eq. 1:

$$ \text{CNR} = \frac{|\text{HU}_{\text{ROI}} - \text{HU}_{\text{SW}}|}{\text{SD}_{\text{SW}}} \quad (1) $$

where $HU_{ROI}$ corresponds to the HU value measured in a ROI placed in the center of one iodine insert. $HU_{SW}$ corresponds to the HU value measured in the ROI of the solid water insert and $SD_{SW}$ is the standard deviation in the same ROI representing the image noise.

For each CT system, contrast, image noise, and CNR values were measured for the four energy levels of VMIs and the five acquisitions. The values obtained for each metric for both CT systems were then compared.

## Potential ICM amount reduction

The variation in CNR according to iodine concentration was determined for VMIs generated by PCCT and EICT at four energy levels.

Assuming the proportional correlation between CNR and the iodine delivery rate as reported by Domenico De Santis et al [38], the potential reduction in iodine was calculated by the rate of reduction in tissue iodine concentration with PCCT while maintaining the same CNR obtained with EICT for the reference concentration. This is illustrated by the following Eq. 2:

$$ \text{Potential iodine reduction (\%)} = \frac{C_{\text{PCCT}} - C_{\text{EICT}}}{C_{\text{EICT}}} \times 100 \quad (2) $$

Where $C_{PCCT}$ are the iodine concentration required with the PCCT to obtain the same CNR value as that measured on the EICT with the reference concentration noted by $C_{EICT}$.

## Statistical analyses

Statistical analyses were performed using version 3.5.1 of R software (R Core Team, 2017, Vienna, Austria). The Shapiro–Wilk test was used to verify the normality of data distribution for HU, noise, and CNR values at different energy levels. As all data were normally distributed (*p*-values > 0.05). Contrast, noise, and CNR values were compared between PCCT and EICT VMIs using the Student *t*-test for all four energy levels. A *p*-value ≤ 0.050 was considered significant.

## Results

### CNR measurement

Table 2 shows the contrast values of each iodine insert according to the energy level for both scanners. HU values for iodine inserts and solid water are shown in Table S1 of the supplementary material. For all energy levels and concentrations ranging from 2 mg/mL to 15 mg/mL, no significant difference was observed between the two CT scanners (*p* > 0.05; Table 2). For 1 mg/mL and all energy levels, contrast values were significantly lower with PCCT with a mean difference of 38.8 ± 3.9% (mean ± standard deviation).

For both scanners, the noise values decreased significantly as the energy level decreased (Fig. 2). For all energy levels, the noise values were significantly lower for PCCT with a mean difference of (49.2 ± 4.3) % at 40 keV, (39.8 ± 5.1)% at 50 keV, (32.5 ± 5.8)% at 60 keV and (25.4 ± 4.0)% at 70 keV.
For both scanners, the CNR values increased significantly as the iodine concentration increased (Fig. 3). For both CT scanners, the CNR values increased significantly as the energy level decreased for iodine concentrations ranging from 2 mg/mL to 15 mg/mL. The opposite pattern was found at 1 mg/mL, except between 60 keV and 70 keV ($p = 0.066$). For iodine concentrations

ranging from 5 mg/mL to 15 mg/mL, the mean CNR values were significantly higher with PCCT than with EICT, at all energy levels ($p < 0.001$). The difference between PCCT and EICT decreased as energy levels increased by $(101.6 \pm 1.6)\%$ at 40 keV, $(69.6 \pm 1.1)\%$ at 50 keV, $(50.6 \pm 0.7)\%$ at 60 keV, and $(34.7 \pm 0.4)\%$ at 70 keV.

With iodine concentration of 2 mg/mL, the mean CNR values were significantly higher with PCCT than with EICT only at energy levels of 40 keV (70.5 ± 18.5%, $p = 0.002$), 50 keV (48.3 ± 17.8%, $p = 0.020$), and 60 keV (36.1 ± 16.5%, $p = 0.040$).

For 1 mg/mL and all energy levels, higher CNR values were found with PCCT than with EICT but the differences were not significant with *p*-values of 0.206, 0.798, 0.606, and 0.356 for 40 keV, 50 keV, 60 keV, and 70 keV, respectively.

<figure><img src="image_8.png" /><figcaption>**Fig. 2** Noise variation on VMIs according to energy level for PCCT and energy-integrating computed tomography (EICT) at a reference radiation dose of 11 mGy</figcaption></figure>

## Potential ICM reduction

Table 3 shows the potential iodine reduction rate according to iodine concentration and energy levels for PCCT, compared with the standard EICT acquisition whilst maintaining the same CNR.

<figure><img src="image_9.png" /><figcaption>**Fig. 3** CNR variation on VMIs according to iodine concentrations for PCCT and EICT: a CNR variation for 40 keV energy level; b CNR variation for 50 keV energy level; c CNR variation for 60 keV energy level; d CNR variation for 70 keV energy level. CNR, Contrast-to-noise ratio</figcaption></figure>
<table><caption>**Table 3** Potential iodine quantity reduction rate according to iodine concentration and energy level for photon counting CT (PCCT) compared with the energy integrating CT (EICT) while maintaining the same CNR</caption><thead><tr><th>Iodine concentration (mg/mL)</th><th>40 keV</th><th>50 keV</th><th>60 keV</th><th>70 keV</th></tr></thead><tbody><tr><td>1</td><td>47%</td><td>33%</td><td>21%</td><td>9%</td></tr><tr><td>2</td><td>48%</td><td>37%</td><td>29%</td><td>24%</td></tr><tr><td>5</td><td>49%</td><td>38%</td><td>30%</td><td>24%</td></tr><tr><td>10</td><td>50%</td><td>39%</td><td>31%</td><td>25%</td></tr><tr><td>15</td><td>50%</td><td>39%</td><td>31%</td><td>26%</td></tr></tbody></table>

The highest reduction potential was found at 40 keV for all iodine concentrations with a mean value of 48.9 ± 1.6%. Potential ICM reduction decreased as the energy level increased, with mean values of 38.5 ± 0.5%, 30.8 ± 0.8% for 50 keV and 60 keV, respectively. For 70 keV, the mean potential reduction of 24.4 ± 1.1% was found for iodine concentrations over 1 mg/mL and a non-significant potential reduction was found for 1 mg/mL.

Discussion

This study demonstrated that for abdomen-pelvis examinations, PCCT provided a lower noise level on VMIs, compared to EICT images. The CNR values were significantly higher with PCCT for all energy levels and iodine concentrations except for 1 mg/mL concentration for which the difference was not significant. A potential iodine reduction rate was found for all iodine concentrations and energy levels except for 1 mg/mL at 70 keV. This potential iodine reduction decreased as the energy level increased.

The results showed an improvement in CNR values with PCCT VMIs compared to EICT. These results are explained by the PCCT's performance in terms of noise reduction on VMIs [22, 45]. In fact, the noise results measured on VMIs showed that the PCCT achieved a reduction ranging from (49.2 ± 4.3)% to (25.4 ± 4.0)% for energy levels from 40 keV to 70 keV, respectively. Further, the CNR values increased linearly with increasing iodine concentration for all energy levels and CT scanners. Similar results were found by De Santis et al [38] and Kok et al [39] on phantoms. Assuming that this linear relationship is also valid for patient images, the improvement in CNR can be exploited to reduce the amount of contrast media with PCCT while maintaining a CNR similar to that obtained with EICT.

For the same energy level of VMIs, the potential reduction in the amount of contrast media was found similar whatever the initial concentration, except for 1 mg/mL at 70 keV, where this potential was lower than for other concentrations. However, this potential decreased as the monoenergetic level of VMIs increased. It should be

noted that the concentrations reported in the literature in the various abdominal structures are higher than 1 mg/mL, with concentrations ranging from 5 mg/mL to 10 mg/mL in the aorta [41, 42] and from 1.23 mg/mL to 4.31 mg/mL in upper abdominal structures [6] or from 1.32 ± 0.33 mg/mL to 2.03 ± 0.42 mg/mL in hepatocellular lesions [43, 44], according to the injection phase (arterial or portal). It is well known that, the monoenergetic level used to interpret VMI for abdominal examinations ranges from 40 keV to 60 keV [15, 46]. Therefore, the potential reduction in the amount of ICM expected for these examinations if the same energy level was used with EICT and PCCT would range from 49 ± 1% at 40 keV to 28 ± 4% at 60 keV energy level. However, the improvement in CNR of VMIs at low energy levels with PCCT offers the possibility of using a lower energy level than with EICT for abdomen imaging. In this case, the VMIs obtained with PCCT offer a higher reduction potential than when using the same energy level as with EICT. This reduction reached 57 ± 2.3% if the energy level used with EICT was 60 keV and that with PCCT was 40 keV.

Transposing these results into clinical application requires vigilance in adapting the injection rate to ensure the injection time is long enough. The injection rate can be calculated from the volume to be injected and the injection time to be respected [47]. The purpose of this adaptation is to maintain a similar distribution of contrast medium in the tissues between acquisition with PCCT and the reference acquisition with EICT. Finally, the results of this study suggest that the improvement in image quality of the VMIs produced by PCCT offers significant prospects to optimize the amount of contrast media injected into patients during abdominal examinations and the amount of contrast agent released into the wastewater, in the years to come.

This study has certain limitations. The first obvious one is that it was a phantom study, limiting the generalizability of results to patients. The CNR values were measured on a phantom with uniform iodine inserts rather than on actual structures of a human abdomen. These results must therefore be confirmed by a study on real patient images. The second limitation concerns the position of the iodine inserts in the center of the phantom; results may differ at the periphery of the phantom. The third is that these acquisition parameters were explored based on our clinical experience with the EICT scanner and the manufacturer's recommended parameters for the PCCT scanner. Finally, the iodine reduction potential is calculated on the basis of a linear relationship between the iodine concentration in the inserts and the CNR. This relationship is demonstrated on phantom, but needs to be verified on patient images by measuring tissue concentration and associated CNR.
In conclusion, for abdomen-pelvis CT protocols, this phantom study showed that using VMIs with PCCT significantly improves the CNR of iodine compared to EICT, offering substantial potential to reduce the amount of required ICM.

**Abbreviations**

CNR Contrast-to-noise ratio

CT Computed tomography

DECT Dual-energy CT

EICT Energy-integrating CT

ICM Iodine contrast media

PCCT Photon-counting CT

ROI Region of interest

VMI Virtual monoenergetic image

**Supplementary information**

The online version contains supplementary material available at https://doi.org/10.1186/s41747-025-00573-2.

**Additional file 1: Table 1 SM:** Mean HU values over a 5 repeated acquisition and associated standard deviation measured for iodine and solid water compared to theoretical values for virtual monochromatic images (VMIs) according to four energy levels for photon-counting CT and energy-integrating CT at standard radiation dose of 11 mGy. **Table 2 SM:** Mean contrast-to-noise (CNR) values over a 5 repeated acquisition and associated standard deviation measured for iodine inserts in virtual monochromatic images (VMIs) according to four energy levels for photon-counting CT and energy-integrating CT at standard radiation dose of 11 mGy.

**Acknowledgements**

We wish to thank Teresa Sawyers, Medical Writer at the BESPIM, Nimes University Hospital, France, for expert editorial assistance. No LLM tools were used to write this manuscript.

**Author contributions**

Conception and design: DD and JG. Administrative support: J-PB. Provision of study materials or patients: DD, JG, SF, and JE. Collection and assembly of data: DD and JG. Data analysis and interpretation: DD and JG. Manuscript writing: all authors. Final approval of manuscript: all authors.

**Funding**

The authors state that this work has not received any funding.

**Data availability**

The datasets analyzed during the current study are available from the corresponding author upon reasonable request.

**Declarations**

**Ethics approval and consent to participate**

Not applicable.

**Consent for publication**

Not applicable.

**Competing interests**

SF, JE, and CC are employees of Siemens Healthineers. The other authors have no relevant conflicts of interest or industry support for the project to declare.

**Author details**

Computed Tomography, Siemens Healthineers AG, Forchheim, Germany.

Received: 4 September 2024 Accepted: 18 February 2025

Published online: 23 March 2025

**References**

1. Schöckel L, Jost G, Seidensticker P et al (2020) Developments in X-ray contrast media and the potential impact on computed tomography. Invest Radiol 55:592–597. https://doi.org/10.1097/RLI.00000000000000696

2. Geenen RWF, Kingma HJ, Van Der Molen AJ (2013) Contrast-induced nephropathy: pharmacology, pathophysiology and prevention. Insights Imaging 4:811–820. https://doi.org/10.1007/s13244-013-0291-3

3. Van Der Molen AJ, Reimer P, Dekkers IA et al (2018) Post-contrast acute kidney injury—part 1: definition, clinical features, incidence, role of contrast medium and risk factors: recommendations for updated ESUR contrast medium safety committee guidelines. Eur Radiol 28:2845–2855. https://doi.org/10.1007/s00330-017-5246-5

4. Van Der Molen AJ, Reimer P, Dekkers IA et al (2018) Post-contrast acute kidney injury. Part 2: risk stratification, role of hydration and other prophylactic measures, patients taking metformin and chronic dialysis patients: recommendations for updated ESUR contrast medium safety committee guidelines. Eur Radiol 28:2856–2869. https://doi.org/10.1007/s00330-017-5247-4

5. Lell M, Kachelrieb M (2023) Computed tomography 2.0: new detector technology, AI, and other developments. Invest Radiol 58:587–601. https://doi.org/10.1097/RLI.00000000000000995

6. Zhang X, Zhang G, Xu L et al (2022) Application of deep learning reconstruction of ultra-low-dose abdominal CT in the diagnosis of renal calculi. Insights Imaging 13:163. https://doi.org/10.1186/s13244-022-01300-w

7. Dabli D, Loisy M, Frandon J et al (2023) Comparison of image quality of two versions of deep-learning image reconstruction algorithm on a rapid kV-switching CT: a phantom study. Eur Radiol Exp 7:1. https://doi.org/10.1186/s41747-022-00314-9

8. Shim J, Kim K, Lee Y (2023) Effect of iodine concentration reduction by comparison of virtual monoenergetic image quality with dual-energy computed tomography. Appl Radiat Isot 200:110967. https://doi.org/10.1016/j.apradiso.2023.110967

9. Greffier J, Villani N, Defez D et al (2023) Spectral CT imaging: technical principles of dual-energy CT and multi-energy photon-counting CT. Diagn Interv Imaging 104:167–177. https://doi.org/10.1016/j.diii.2022.11.003

10. Dabli D, Frandon J, Belaouni A et al (2022) Optimization of image quality and accuracy of low iodine concentration quantification as function of dose level and reconstruction algorithm for abdominal imaging using dual-source CT: a phantom study. Diagn Interv Imaging 103:31–40. https://doi.org/10.1016/j.diii.2021.08.004

11. Kojima T, Shirasaka T, Kondo M et al (2021) A novel fast kilovoltage switching dual-energy CT with deep learning: accuracy of CT number on virtual monochromatic imaging and iodine quantification. Phys Med 81:253–261. https://doi.org/10.1016/j.ejmp.2020.12.018

12. Doerner J, Hauger M, Hickethier T et al (2017) Image quality evaluation of dual-layer spectral detector CT of the chest and comparison with conventional CT imaging. Eur J Radiol 93:52–58. https://doi.org/10.1016/j.ejrad.2017.05.016

13. Zhang D, Li X, Liu B (2011) Objective characterization of GE discovery CT750 HD scanner: gemstone spectral imaging mode. Med Phys 38:1178–1188. https://doi.org/10.1118/1.3551999

14. McCollough CH, Boedeker K, Cody D et al (2020) Principles and applications of multienergy CT: report of AAPM task group 291. Med Phys. https://doi.org/10.1002/mp.14157

15. D'Angelo T, Cicero G, Mazziotti S et al (2019) Dual energy computed tomography virtual monoenergetic imaging: technique and clinical applications. Br J Radiol. https://doi.org/10.1259/bjr.20180546

16. Patel BN, Farjat A, Schabel C et al (2018) Energy-specific optimization of attenuation thresholds for low-energy virtual monoenergetic images in renal lesion evaluation. AJR Am J Roentgenol 210:W205–W217. https://doi.org/10.2214/AJR.17.18641

<footnote n="1">¹Department of medical imaging, CHU Nîmes, Univ Montpellier, Nîmes Medical Imaging Group, UR UM103 IMAGINE, Nîmes, France. ²Department of</footnote>

<footnote n="3">³MIPA, University of Nîmes, Nîmes, France.</footnote>
17. Lee SM, Kim SH, Ahn SJ et al (2018) Virtual monoenergetic dual-layer, dual-energy CT enterography: optimization of keV settings and its added value for Crohn's disease. Eur Radiol 28:2525-2534. https://doi.org/10.1007/s00330-017-5215-z

18. Martin SS, Pfeifer S, Wichmann JL et al (2017) Noise-optimized virtual monoenergetic dual-energy computed tomography: optimization of kilo-electron volt settings in patients with gastrointestinal stromal tumors. Abdom Radiol (NY) 42:718-726. https://doi.org/10.1007/s00261-016-1011-5

19. Sakabe D, Nakaura T, Oda S et al (2020) Decreasing the radiation dose for contrast-enhanced abdominal spectral CT with a half contrast dose: a matched-pair comparison with a 120 kVp protocol. BJR Open 2:20200006. https://doi.org/10.1259/bjro.20200006

20. He J, Wang Q, Ma X, Sun Z (2015) Dual-energy CT angiography of abdomen with routine concentration contrast agent in comparison with conventional single-energy CT with high concentration contrast agent. Eur J Radiol 84:221-227. https://doi.org/10.1016/j.ejrad.2014.11.025

21. Van Hamersvelt RW, Eijsvoogel NG, Mihl C et al (2018) Contrast agent concentration optimization in CTA using low tube voltage and dual-energy CT in multiple vendors: a phantom study. Int J Cardiovasc Imaging 34:1265-1275. https://doi.org/10.1007/s10554-018-1329-x

22. Flohr T, Schmidt B (2023) Technical basics and clinical benefits of photon-counting CT. Invest Radiol 58:441-450. https://doi.org/10.1097/RLI.00000000000098028

23. Danielsson M, Persson M, Sjölin M (2021) Photon-counting x-ray detectors for CT. Phys Med Biol 66:03TR01. https://doi.org/10.1088/1361-6560/abc5a5

24. Willemink MJ, Persson M, Pourmorteza A et al (2018) Photon-counting CT: technical principles and clinical prospects. Radiology 289:293-312. https://doi.org/10.1148/radiol.2018172656

25. Si-Mohamed SA, Mialhes J, Rodesch P-A et al (2021) Spectral photon-counting CT technology in chest imaging. J Clin Med 10:5757. https://doi.org/10.3390/jcm10245757

26. Greffier J, Faby S, Pastor M et al (2024) Comparison of low-energy virtual monoenergetic images between photon-counting CT and energy-integrating detectors CT: a phantom study. Diagn Interv Imaging S2211568424000445. https://doi.org/10.1016/j.diii.2024.02.009

27. Schwartz FR, Samei E, Marin D (2023) Exploiting the potential of photon-counting CT in abdominal imaging. Invest Radiol 58:488-498. https://doi.org/10.1097/RLI.00000000000000949

28. Higashigaito K, Euler A, Eberhard M et al (2022) Contrast-enhanced abdominal CT with clinical photon-counting detector CT: assessment of image quality and comparison with energy-integrating detector CT. Acad Radiol 29:689-697. https://doi.org/10.1016/j.acra.2021.06.018

29. Zhan X, Zhang R, Niu X et al (2023) Comprehensive evaluations of a prototype full field-of-view photon counting CT system through phantom studies. Phys Med Biol 68:175007. https://doi.org/10.1088/1361-6560/acebb3

30. Si-Mohamed SA, Boccalini S, Villien M et al (2023) First experience with a whole-body spectral photon-counting CT clinical prototype. Invest Radiol 58:459-471. https://doi.org/10.1097/RLI.00000000000000965

31. Almqvist H, Crotty D, Nyren S et al (2024) Initial clinical images from a second-generation prototype silicon-based photon-counting computed tomography system. Acad Radiol 31:572-581. https://doi.org/10.1016/j.acra.2023.06.031

32. Saeed S, Niehoff JH, Boriesosdick J et al (2024) Minimizing contrast media dose in CT pulmonary angiography with clinical photon counting using high pitch technique. Acad Radiol 31:686-692. https://doi.org/10.1016/j.acra.2023.05.018

33. Emrich T, O'Doherty J, Schoepf UJ et al (2023) Reduced iodinated contrast media administration in coronary CT angiography on a clinical photon-counting detector CT system: a phantom study using a dynamic circulation model. Invest Radiol 58:148-155. https://doi.org/10.1097/RLI.00000000000000911

34. Rajiah PS, Dunning CAS, Rajendran K et al (2023) High-pitch multienergy coronary CT angiography in dual-source photon-counting detector CT scanner at low iodinated contrast dose. Invest Radiol 58:681-690. https://doi.org/10.1097/RLI.0000000000000961

35. Cundari G, Deilmann P, Mergen V et al (2024) Saving contrast media in coronary CT angiography with photon-counting detector CT. Acad Radiol 31:212-220. https://doi.org/10.1016/j.acra.2023.06.025

36. Higashigaito K, Mergen V, Eberhard M et al (2023) CT angiography of the aorta using photon-counting detector CT with reduced contrast media volume. Radiol Cardiothorac Imaging 5:e220140. https://doi.org/10.1148/ryct.220140. 26

37. Rau S, Soschynski M, Schlett CL et al (2023) Spectral aortoiliac photon-counting CT angiography with minimal quantity of contrast agent. Radiol Case Rep 18:2180-2182. https://doi.org/10.1016/j.radcr.2023.01.066. 11

38. De Santis D, Caruso D, Schoepf UJ et al (2018) Contrast media injection protocol optimization for dual-energy coronary CT angiography: results from a circulation phantom. Eur Radiol 28:3473-3481. https://doi.org/10.1007/s00330-018-5308-3

39. Kok M, Mihl C, Hendriks BM et al (2016) Optimizing contrast media application in coronary CT angiography at lower tube voltage: evaluation in a circulation phantom and sixty patients. Eur J Radiol 85:1068-1074. https://doi.org/10.1016/j.ejrad.2016.03.022

40. Corrias G, Sawan P, Mahmood U et al (2019) Dual energy computed tomography analysis in cancer patients: What factors affect iodine concentration in contrast enhanced studies? Eur J Radiol 120:108698. https://doi.org/10.1016/j.ejrad.2019.108698

41. Nehnahi M, Simon G, Moinet R et al (2023) Quantifying iodine concentration in the normal bowel wall using dual-energy CT: influence of patient and contrast characteristics. Sci Rep 13:22714. https://doi.org/10.1038/s41598-023-50238-6

42. Patel BN, Vernuccio F, Meyer M et al (2019) Dual-energy CT material density iodine quantification for distinguishing vascular from nonvascular renal lesions: normalization reduces intermanufacturer threshold variability. AJR Am J Roentgenol 212:366-376. https://doi.org/10.2214/AJR.18.20115

43. Mo H, Huang R, Wei X et al (2023) Diagnosis of metastatic lymph lodes in patients with hepatocellular carcinoma using dual-energy computed tomography. J Comput Assist Tomogr 47:355-360. https://doi.org/10.1097/RCT.00000000000001429

44. Patel BN, Rosenberg M, Vernuccio F et al (2018) Characterization of small incidental indeterminate hypoattenuating hepatic lesions: added value of single-phase contrast-enhanced dual-energy CT material attenuation analysis. AJR Am J Roentgenol 211:571-579. https://doi.org/10.2214/AJR.17.19170

45. Greffier J, Faby S, Pastor M et al (2024) Comparison of low-energy virtual monoenergetic images between photon-counting CT and energy-integrating detectors CT: a phantom study. Diagn Interv Imaging 105:311-318. https://doi.org/10.1016/j.diii.2024.02.009

46. Lahuna L, Greffier J, Goupil J et al (2023) Determining the optimal energy level of virtual monoenergetic images in dual-source CT for diagnosis of bowel obstruction and colitis. Diagnostics (Basel) 13:3491. https://doi.org/10.3390/diagnostics13233491

47. Caruso D, Eid M, Schoepf UJ et al (2017) Optimizing contrast media injection protocols in computed tomography angiography at different tube voltages: evaluation in a circulation phantom. J Comput Assist Tomogr 41:804-810

**Publisher's Note**

Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.