Contain Nonbinding Recommendations
Submission of Premarket Notifications 
for Magnetic Resonance Diagnostic 
Devices
Guidance for Industry and Food and 
Drug Administration Staff
Document issued on: October 10, 2023
Document originally issued on November 18, 2016
This document supersedes Submission of Premarket Notifications for Magnetic 
Resonance Diagnostic Devices, Guidance for Industry and Food and Drug 
Administration Staff, issued December 22, 2016
For questions about this document, contact Office of Health Technology 8 (OHT8): Office of
Radiological Health at RadHealth@fda.hhs.gov.
U.S. Department of Health and Human Services
Food and Drug Administration
Center for Devices and Radiological Health Contain Nonbinding Recommendations
Preface
Public Comment
You may submit electronic comments and suggestions at any time for Agency consideration to 
http://www.regulations.gov. Submit written comments to the Dockets Management Staff, Food 
and Drug Administration, 5630 Fishers Lane, Room 1061, (HFA-305), Rockville, MD 20852-
1740. Identify all comments with the docket number FDA-2015-D-2148. Comments may not be 
acted upon by the Agency until the document is next revised or updated.
Additional Copies
Additional copies are available from the Internet. You may also send an email request to CDRH-
Guidance@fda.hhs.gov to receive a copy of the guidance. Please use the document number 
GUI00000340 and complete title of the guidance in the request. Contain Nonbinding Recommendations
Table of Contents
I.  Introduction ............................................................................................................................. 1
II.  Scope ....................................................................................................................................... 2
III.  Relevant Standards.................................................................................................................. 3
A.  NEMA Standards ............................................................................................................ 3
B.  IEC 60601-2-33 .............................................................................................................. 3
C.  Other Applicable Standards ............................................................................................ 4
IV.  Describing Your Device in a 510(k) Premarket Notification ................................................. 4
A.  Indications for Use .......................................................................................................... 4
B.  Device Description.......................................................................................................... 5
V.  Electrical, Mechanical, Structural, and Related System Safety .............................................. 8
VI.  Physical Laboratory Testing ................................................................................................... 9
A.  Performance .................................................................................................................... 9
B.  Safety ............................................................................................................................ 10
VII. Clinical Images ..................................................................................................................... 11
VIII. Device Modifications ............................................................................................................ 12
IX.  Labeling ................................................................................................................................ 14
A.  RF Coil Labeling........................................................................................................... 14
B.  Summary Specification Sheet ....................................................................................... 14
C.  User Manual .................................................................................................................. 15
D.  Site Planning Information ............................................................................................. 17 Contain Nonbinding Recommendations
Submission of Premarket Notifications 
for Magnetic Resonance Diagnostic 
Devices
Guidance for Industry and Food and 
Drug Administration Staff
This guidance represents the current thinking of the Food and Drug Administration (FDA or 
Agency) on this topic. It does not establish any rights for any person and is not binding on FDA or 
the public. You can use an alternative approach if it satisfies the requirements of the applicable 
statutes and regulations. To discuss an alternative approach, contact the FDA staff or Office 
responsible for this guidance as listed on the title page. 
I. Introduction 
 
This guidance document provides a detailed description of the information that should be included in 
a premarket notification for a magnetic resonance diagnostic device (MRDD). This document is 
intended to be used in conjunction with information regarding the content and format of a 510(k) 
premarket notification1. The approach outlined in this guidance document is intended to facilitate the 
timely review and marketing clearance of MRDDs.
The guidance reflects an update to harmonize with the 4th edition of the FDA recognized standard 
IEC 60601-2-33 Medical electrical equipment - Part 2-33: Particular requirements for the basic 
safety and essential performance of magnetic resonance equipment for medical diagnosis. In the 4th 
edition of IEC 60601-2-33, the main magnetic field (B0) hazard area was re-defined as anywhere 
where the magnetic field is equal to or greater than 0.9 mT as the space around MR equipment where 
the static magnetic field can cause harm. Prior versions of IEC 60601-2-33 had defined this 
controlled access area as 0.5 mT. For the current edition of the FDA-recognized consensus 
standard(s) referenced in this document, see theFDA Recognized Consensus Standards Database.2 If 
submitting a Declaration of Conformity to a recognized standard, we recommend you include the 
appropriate supporting documentation. For more information regarding use of consensus standards in 
1 Refer to FDA guidance titled Electronic Submission Template for Medical Device 510(k) Submissions, available at 
https://www.fda.gov/regulatory-information/search-fda-guidance-documents/electronic-submission-template-medical-
device-510k-submissions
2Available at https://www.accessdata.fda.gov/scripts/cdrh/cfdocs/cfStandards/search.cfm
1 Contain Nonbinding Recommendations
regulatory submissions, refer to the FDA guidance titled Appropriate Use of Voluntary Consensus 
Standards in Premarket Submissions for Medical Devices.3
MRDDs are also electronic products under section 531(2) of Subchapter C (Electronic Product 
Radiation Control (EPRC)) of the Federal Food, Drug, and Cosmetic Act (FD&C Act). As such, 
MRDDs are subject to the radiological health requirements in Title 21, Subchapter J, Parts 1000 
through 1050 of the Code of Federal Regulations, including applicability of general and specific 
performance standards (21 CFR Parts 1010-1050) and other general requirements for reporting and 
recordkeeping (21 CFR Part 1002), notification and corrective actions for defective or non-compliant 
electronic products (21 CFR Parts 1003 and 1004), and importation (21 CFR Part 1005). 
In general, FDAs guidance documents do not establish legally enforceable responsibilities. Instead, 
guidances describe the Agencys current thinking on a topic and should be viewed only as 
recommendations, unless specific regulatory or statutory requirements are cited. The use of the word 
should in Agency guidances means that something is suggested or recommended, but not required.
II. Scope 
 
This document is applicable to MRDDs as defined in 21 CFR 892.1000:  
21 CFR 892.1000: Magnetic resonance diagnostic device.
(a) Identification. A magnetic resonance diagnostic device is intended for general diagnostic 
use to present images which reflect the spatial distribution and/or magnetic resonance spectra 
which reflect frequency and distribution of nuclei exhibiting nuclear magnetic resonance. 
Other physical parameters derived from the images and/or spectra may also be produced. The 
device includes hydrogen-1 (proton) imaging, sodium-23 imaging, hydrogen-1 spectroscopy, 
phosphorus-31 spectroscopy, and chemical shift imaging (preserving simultaneous frequency 
and spatial information).
(b) Classification. Class II. 
MRDDs are Class II medical devices that require premarket notification and an agency determination 
of substantial equivalence prior to marketing. Three product codes are currently used to identify these 
devices:
LNH  Nuclear Magnetic Resonance Imaging System
LNI  Nuclear Magnetic Resonance Spectroscopic System
MOS - Magnetic Resonance Specialty Coil
The principal components of current MRDDs include the main magnet, shim and gradient systems, 
radiofrequency transmitter and receiver, transmit and receive coils, power supplies, computer and 
software, patient supports, and physiological gating devices. 
3 Available at https://www.fda.gov/regulatory-information/search-fda-guidance-documents/appropriate-use-voluntary-
consensus-standards-premarket-submissions-medical-devices
2 Contain Nonbinding Recommendations
This guidance document is applicable to premarket notifications for magnetic resonance imaging 
(MRI) and magnetic resonance spectroscopy (MRS) systems, components, and accessories, and 
modifications to systems, components, and accessories, which could significantly affect the safety or 
effectiveness of the MRDD and trigger the need for premarket notification submission prior to 
marketing. 
The information in this guidance document is also applicable to the MRI system components of dual-
modality devices, such as PET/MRI systems. 
III. Relevant Standards  
 
Standards may be used only when applicable (section 514(c)(1)(A) of the FD&C Act); not all 
standards specified below are applicable to all MRDD 510(k) submissions.
A. NEMA Standards 
 
Standards promulgated by the National Electrical Manufacturers Association (NEMA) provide 
standardized test methods for the assessment of performance and safety parameters for MRDDs. The 
NEMA standards only prescribe standard measurement methods and do not specify acceptance 
criteria. You should define acceptance criteria for your device. FDA will evaluate acceptance criteria 
on a case-by-case basis depending on the intended use and specific technological characteristics of 
your device. NEMA test methods recognized by FDA include:  
 MS 1 Determination of Signal-to-Noise Ratio (SNR) in Diagnostic Magnetic Resonance 
Images
 MS 2 Determination of Two-dimensional Geometric Distortion in Diagnostic Magnetic 
Resonance Images
 MS 3 Determination of Image Uniformity in Diagnostic Magnetic Resonance Images
 MS 4 Acoustic Noise Measurement Procedure for Diagnostic Magnetic Resonance Imaging 
Devices
 MS 5 Determination of Slice Thickness in Diagnostic Magnetic Resonance Imaging
 MS 6 Determination of Signal-to-Noise Ratio and Image Uniformity for Single-Channel, 
Non-Volume Coils in Diagnostic Magnetic Resonance Imaging (MRI)
 MS 8 Characterization of the Specific Absorption Rate (SAR) for Magnetic Resonance 
Imaging Systems
 MS 9 Characterization of Phased Array Coils for Diagnostic Magnetic Resonance Images
 MS 10 Determination of Local Specific Absorption Rate (SAR) in Diagnostic Magnetic 
Resonance Imaging
 MS 12 Quantification and Mapping of Geometric Distortion for Special Applications
B. IEC 60601233 
 
3 Contain Nonbinding Recommendations
The International Electrotechnical Commission (IEC) standard, IEC 60601-2-33 Medical Electrical 
equipment  Part 2-33: Particular requirements for the basic safety and essential performance of 
magnetic resonance equipment for medical diagnosis, is the international standard for the safety of 
magnetic resonance equipment intended for medical diagnosis. IEC 60601-2-33 provides particular 
specifications for magnetic resonance diagnostic equipment and takes precedence over the 
specifications provided in the general IEC 60601 series. 
The NEMA standards for measuring acoustic noise (NEMA MS 4) and SAR (NEMA MS 8) have 
been incorporated into IEC 60601-2-33. However, IEC 60601-2-33 does not address performance 
issues, such as SNR, image uniformity, geometric distortion and slice thickness. 
C. Other Applicable Standards 
 
  ISO 10993-1 Biological evaluation of medical devices  Part 1: Evaluation and Testing 
within a Risk Management Process. This standard applies to patient-contacting materials in 
MRDDs.
 NEMA PS 3.1 - 3.20 Digital Imaging and Communications in Medicine (DICOM). This 
standard specifies formats for the digital exchange of medical images.
 ANSI/AAMI ES60601-1 Medical electrical equipment - Part 1: General Requirements for 
Basic Safety and Essential Performance
 IEC 60601-1 Medical electrical equipment - Part 1: General requirements for basic safety 
and essential performance (- Note: This standard is recognized with relevant U.S. national 
differences applied)
 IEC 60601-1-2 Medical electrical equipment - Part 1-2: General Requirements for Basic 
Safety and Essential Performance - Collateral standard: Electromagnetic Disturbances - 
Requirements and tests
IV. Describing Your Device in a 510(k) Premarket Notification 
 
When submitting a 510(k) premarket notification for a MRDD, you must include the information 
required under 21 CFR 807.87. You should identify your device by regulation and product code and 
should also include the information below.  
 
A.  Indications for Use 
 
Youshould describe with particularity, as described below, the Indications for Use (IFU) for your 
device. The device labeling, training materials, performance claims, promotional materials, and any 
other materials in the submission should all be consistent with the IFU. 
 
Specific clinical indications (for example, disease identification or rule-out, diagnosis or prognosis 
with respect to disease staging or severity, and prevention or reduction in morbidity and/or mortality 
associated with particular diseases) are beyond the scope of this document. While the Agency 
handles each such application on an individual basis, the Agency believes that clinical studies are
4 Contain Nonbinding Recommendations
often necessary to support such specific clinical indications. Prior to the submission of any such 
application, you are encouraged to contact the Agency to discuss the level of supporting evidence 
required and clinical trial study design. For information on FDAs Q-submission process, refer to 
FDA guidance titled Requests for Feedback and Meetings for Medical Device Submissions: The Q-
Submission Program.4
MRDDs often contain protocols recommended for specific applications (for example, adult abdomen, 
pediatric brain, etc.). For MRDDs, these specific protocols do not necessarily create a new specific 
indication for the MRDD, provided that disease-specific or diagnostic claims are not made. For 
example, a Pediatric Brain protocol would be acceptable under a general IFU, but a Pediatric 
Epilepsy Foci identification protocol would likely be interpreted as a new specific indication.
The Indications for Use for radiofrequency (RF) coils and accessory devices should specify the MRI 
system(s) with which the devices are intended to be used. The level of detail necessary will depend 
upon the individual device and the MRI systems with which it is intended to be used. For example, 
FDA generally believes that specifying the manufacturer and field strength of the compatible MRI 
system is appropriate for an RF coil.   
B. Device Description 
 
You should provide a comprehensive description of your device in the premarket notification that 
includes the information below: 
(1) Magnet  
A full description of the main magnet including:
 Field strength and type of magnet (superconducting, resistive or permanent)
 Dimensions of the patient-accessible bore space
 Type of installation (fixed, mobile, interventional, or transportable)
 Design characteristics of the magnet, including weight, cryogens and boil-off rates (if 
applicable), bore dimensions, type of shielding, shimming method
 Performance characteristics of the magnet, including decay characteristics of the magnetic 
field in the event of a quench, fringe field maps (including 0.5 mT, 0.9 mT, 3 mT, 5 mT, 10 
mT, 20 mT, 40 mT and 200 mT contours), stability of the field (ppm/hr) over a prolonged 
period of time, and spatial homogeneity (ppm/volume)
(2) Gradient System  
A full description of the gradient system including:
 A dimensioned illustration of the gradient tube 
 Information on shielding and cooling 
 Maximum gradient amplitude (per axis) in T/m, rise time (ms), and maximum slew rate 
(T/m/s) 
4Available at https://www.fda.gov/regulatory-information/search-fda-guidance-documents/requests-feedback-and-
meetings-medical-device-submissions-q-submission-program
5 Contain Nonbinding Recommendations
 A description of how cardiac and peripheral nerve stimulation control is implemented, 
including a discussion of any hardware or software mechanism to limit gradient output 
(dB/dt) for systems capable of exceeding the time rate of change of the magnetic field (dB/dt) 
of 20 T/s.  
 
If peripheral nerve stimulation (PNS) is possible, you should describe the level of gradient 
output at which the operator is notified. You should also include information about how the 
operator is informed that PNS is possible. Painful stimulation should be avoided. 
 You should provide information about the operating modes implemented on the system (for 
example, Normal, First Level Controlled, Second Level Controlled) and how users navigate 
between the different modes.
(3) Radiofrequency System  
You should provide a description of the architecture of the RF transmit-receive system, including the 
number of transmit and receive channels, amplifier peak power and duty cycle.  
 
FDA recommends that all MRDDs retain the ability to operate in quadrature transmit mode. If 
applicable, you should include a description of how the user identifies and selects quadrature transmit 
mode on your system.
(4) RF Coils  
For each RF coil included with your system, you should provide the following information:
 Type of coil (transmit, receive, transmit/receive)  
 Description of the hardware characteristics of the coil (for example, geometry, materials, 
dimensions)
 A description of the coil design (for example, linear, quadrature, phased array, multi-channel 
transmit) 
 Intended use (resonant nuclei, frequencies, anatomical region of interest)
 Schematic of the coil design including the location of individual coil elements
 Circuit diagrams
 A description of the decoupling method(s) employed (for receive-only coils) 
(5) SAR Management and Control System 
  A description of the SAR management and control system, including how whole-body 
averaged (avg-WB), partial body (PB), head-averaged, local (10g-averaged) SAR, and 
specific absorbed energy (i.e., SAR over the examination time) control is implemented 
 Information about the operating modes implemented on the system and how users navigate 
between the different operating modes
 The specification for accuracy and uncertainty in the console-reported SAR values  
 A description of how energy deposition information is communicated to the user
 You should explain all warnings or other feedback provided to the user
(6) Pulse Sequences  
For purposes of this guidance, FDA uses the term pulse sequence to mean a technique like spin 
echo, gradient echo, echo planar imaging, etc., including their parameterization (for example echo 
6 Contain Nonbinding Recommendations
time, repetition time, number of slices). For each pulse sequence provided on your MRDD, you 
should provide the following:
 Pulse sequence diagram 
 Pulse sequence type (for example spin echo, gradient echo, fast spin echo, 2D/3D)
 Contrast characteristics (for example, T1, T2, weighting, fat saturation)
 k-space trajectory (spiral, Cartesian, etc.)
 Associated options (shimming, parallel imaging, saturation pulses, etc.) 
 Coil preference or limitations (if any)
 Additional accessory equipment required (for example, respiratory and/or cardiac gating, 
elastography drivers)
 Summary of image reconstruction method (for example FFT, compressed sensing)
(7) Imaging Protocols  
For purposes of this guidance, FDA uses the term imaging protocol to mean a workflow tool 
consisting of multiple pulse sequences prescribed in a defined order. For each manufacturer-provided 
protocol, you should provide the information below. 
 Target anatomy
 A list of the pulse sequences included in each protocol 
 Coil preference or restriction (if any) 
 Whether the protocol is intended to be used in combination with exogenous contrast media 
(8) Image Processing  
For purposes of this guidance, FDA uses the term image processing to mean algorithms operating 
on acquired image data. For the purposes of this guidance, FDA does not consider reconstruction 
algorithms to be image processing. For each image processing feature offered with the MRDD, you 
should provide:
 Inputs, their data formats, and methods of input (for example, feed from other modules, 
manual input) 
 Functional description of the algorithm(s)
 Level of user interaction (for example, automated, semi-automated and manual, whether 
results can be edited or need to be reviewed by the user)
 Outputs, their data formats, and how they are displayed
Software - In general, a 510(k) application should include software documentation consistent 
with a Basic Documentation Level as specified in the FDA guidance titled Content of Premarket 
Submissions for Device Software Functions.5 The actual Documentation Level for your device 
may vary based on the specifics of your device.
If the device meets the definition of a cyber device under section 524B(c) of the FD&C Act, 
cybersecurity documentation under section 524B(b) of the FD&C Act is required as a part of the 
premarket submission. For more information on this topic, refer to FDA guidance titled 
5Available at https://www.fda.gov/regulatory-information/search-fda-guidance-documents/content-premarket-
submissions-device-software-functions
7 Contain Nonbinding Recommendations
Cybersecurity in Medical Devices: Quality System Considerations and Content of Premarket 
Submissions6.
If your device includes off-the-shelf software, you should provide the additional information as 
recommended in the FDA guidance titled Off-The-Shelf Software Use in Medical Devices and 
FDA guidance titled Cybersecurity for Networked Medical Devices Containing Off-The-Shelf 
(OTS) Software.7
(9) Accessories  
Your 510(k) submission should include a list of the accessories provided with the system (for 
example, physiological monitoring accessories such as EKG leads, pulse oximeters, respiratory 
and/or cardiac gating, elastography drivers, or patient positioners).
(10) Additional Information  
  A description of the patient table including dimensions, positioning accuracy and maximum 
supported weight
 Information about patient communication equipment, such as the nurse call or panic button
 Information about recommended RF shielding and how the Special Environment 
specifications in IEC 60601-1-2 are implemented and how integrity is maintained during 
operation
 FDA encourages the adoption of the Fixed Parameter Option on your MRDD. If fixed 
parameter options (such as FPO:B) are implemented on your system, you should describe the 
implementation. For additional information on a Fixed Parameter Option, see IEC 60601-2-
33.
V. Electrical, Mechanical, Structural, and Related System 
Safety 
You should evaluate the safety aspects of your device according to the following FDA-recognized 
consensus standards:
 ANSI/AAMI ES60601-1 Medical electrical equipment - Part 1: General requirements for 
basic safety and essential performance
 IEC 60601-1 Medical electrical equipment - Part 1: General requirements for basic safety 
and essential performance (- Note: This standard is recognized with relevant U.S. national 
differences applied)
 IEC 60601-1-2 Medical electrical equipment - Part 1-2: General requirements for basic 
safety and essential performance - collateral standard: electromagnetic disturbances - 
requirements and tests
6Available at https://www.fda.gov/regulatory-information/search-fda-guidance-documents/content-premarket-
submissions-management-cybersecurity-medical-devices
7Available at https://www.fda.gov/regulatory-information/search-fda-guidance-documents/cybersecurity-networked-
medical-devices-containing-shelf-ots-software
8 Contain Nonbinding Recommendations
For additional information on providing EMC information in a premarket submission, please refer to 
the FDA guidance titled Electromagnetic Compatibility (EMC) of Medical Devices8.
If your MRDD incorporates wireless technology for data communication meeting the IEC 60601-1-2 
standard is insufficient to demonstrate that the wireless technology will not be susceptible to 
electromagnetic interferences and continue to perform as intended. For additional information, refer 
to the FDA guidance titled Radio-Frequency Wireless Technology in Medical Devices9. 
VI. Physical Laboratory Testing 
To demonstrate the substantial equivalence of your MRDD, you should provide the performance 
testing specified below. A number of the tests specified below can be performed in accordance with 
FDA-recognized consensus standards. 
Not all of the testing below is applicable to all accessories. You should perform testing of relevant 
characteristics that may affect safety and effectiveness. 
A. Performance 
 
To demonstrate the performance of your MRDD, you should include an assessment of image quality 
metrics (such as signal-to-noise ratio, geometric distortion, image uniformity, slice thickness, and 
spatial resolution), as well as spectroscopy performance. The measurement method used should be 
specified for all testing, including for the performance metrics outlined below:   
(1) Imaging  
  Signal to Noise Ratio (SNR)  The measured SNR value as well as pre-determined pass/fail 
acceptance criteria for all coils included in the scope of the submission. 
 Geometric Distortion  Distortion measured in all three slice orientations (sagittal, coronal 
and transverse) as well as pre-determined pass/fail acceptance criteria. 
 Image Uniformity  Image uniformity measurements and/or gray-scale uniformity maps, as 
well as pre-determined pass/fail acceptance criteria for all coils included in the scope of the 
submission.
 Slice Thickness  Full-width-half-maximum (FWHM) values as well as pre-determined 
pass/fail acceptance criteria. 
 Spatial Resolution  High contrast spatial resolution of the system demonstrated using 
suitable phantoms for the clinical pulse sequence protocols using the smallest field of view 
(FOV). 
8Available at https://www.fda.gov/regulatory-information/search-fda-guidance-documents/electromagnetic-
compatibility-emc-medical-devices
9Available at https://www.fda.gov/regulatory-information/search-fda-guidance-documents/radio-frequency-wireless-
technology-medical-devices-guidance-industry-and-fda-staff
9 Contain Nonbinding Recommendations
 Image contrast validation  The image contrast behavior for new pulse sequences. For 
example, fat saturation pulse sequences should demonstrate adequate fat signal suppression in 
a phantom composed of fat and water. 
(2) Spectroscopy  
No standardized tests have been developed for magnetic resonance spectroscopy performance. FDA 
recommends the following performance testing for systems with spectroscopy scan protocols. All test 
results should be accompanied by a description of the test methods used, including the pulse 
sequences and coils utilized, and the geometry and composition of the phantoms. The target 
anatomical region and the RF hardware used should be specified. Phantom testing should include 
performance characteristic such as: 
  Spatial Localization Accuracy  Comparison of desired and actual volume
 Spectral Resolution  Full-width-half-maximum of the water resonance using the clinical 
protocols (for example, single voxel or chemical shift imaging)
 Signal to Noise Ratio  Ratio of peak amplitude to standard deviation of background for key 
metabolites (for example, N-Acetyl aspartate or lactate)
 Water suppression  Ratio of area of water peak with and without suppression
 Decoupling  Comparison of SNR with and without decoupling
 Spectral Data Processing  Validation of spectral post-processing techniques 
B. Safety 
 
To demonstrate the safety of your MRDD, you should address acoustic noise, gradient-induced 
electric fields, RF energy deposition and biocompatibility and flammability of patient-contacting 
materials as outlined below. 
(1) Acoustic Noise 
You should measure the acoustic output of your system and specify the measurement method used 
(for example, maximum gradient acoustic noise or maximum clinical acoustic noise). You should 
report unweighted peak sound pressure level (Lpeak) and the time integral of the A-weighted sound 
pressure level (L eq). 
A
(2) Gradient-induced Nerve Stimulation 
For gradient systems capable of producing dB/dt greater than 20 T/s, you should conduct volunteer 
studies with a minimum of 11 subjects to determine the threshold value for mild and painful 
peripheral nerve stimulation. If painful stimulation can be induced in your system, you should also 
determine the threshold of painful stimulation and provide an explanation of how painful stimulation 
is avoided.
(3) RF Energy Deposition  
For volume-transmit coils, you should measure whole-body averaged, head averaged, and/or partial 
body SAR values. You should specify the test method used (for example, pulse energy or 
calorimetry). You should report both the measured and the scanner-displayed SAR values and should 
specify the uncertainty boundaries for all reported SAR values.
10 Contain Nonbinding Recommendations
For surface transmit coils, you should measure local 10g-averaged SAR values. You should report 
both the measured and the scanner-displayed SAR values and include uncertainty boundaries for all 
reported SAR values. 
For multi-channel transmit coils, you should provide a discussion of how the peak local (10g-
averaged) SAR values in your system compare to peak values in quadrature volume coils that 
conform to the current whole-body and whole-head IEC 60601-2-33 SAR limits. This discussion 
should encompass the entire patient population and anatomical scan landmarks for which your device 
is indicated. If computational models are used to support substantial equivalence, these models 
should be accompanied by validation and uncertainty data. You should include uncertainty 
boundaries for all reported SAR values.
(4) Surface Heating of RF Receive Coils  
To ensure patient safety and prevent thermal injury to patients undergoing MR exams, you should 
measure the temperature rise of all receive-only coils included with the system. The results reported 
to FDA should include an assessment of why the measured temperature rise is acceptable and does 
not pose a risk to patients. FDA recommends that temperature be measured at locations in the coil 
pre-determined to be the local hot spots, and that you conduct this test for the coil in the normal 
operating condition, and for the single fault condition of the coil left in the bore of the magnet 
unplugged. 
(5) Biocompatibility  
You should address biocompatibility for any patient-contacting materials. For additional information, 
refer to FDA guidance titled Use of International Standard ISO 10993-1 Biological evaluation of 
medical devices  Part 1: Evaluation and testing within a risk management process10. 
(6) Fixed parameter options  
If fixed parameter options (such as FPO: B) are implemented on your system, you should 
demonstrate that your device operates within the FPO limits.  
 
VII. Clinical Images 
 
You should provide sample clinical images to support the ability of your system to generate 
diagnostic quality images. 
For newly introduced systems, you should provide sample clinical images for all pulse sequences. 
If a pulse sequence is employed in multiple clinical protocols, the sample clinical images should 
support the ability of the pulse sequence to achieve the desired contrast characteristics in those 
clinical protocols. Scientific rationale and a limited set of sample clinical images may be adequate to 
support the use of a pulse sequence across multiple protocols.
10Available at https://www.fda.gov/regulatory-information/search-fda-guidance-documents/use-international-standard-
iso-10993-1-biological-evaluation-medical-devices-part-1-evaluation-and
11 Contain Nonbinding Recommendations
You should provide images to the Agency in electronic DICOM format. You should remove any 
patient identifiers prior to submitting images to FDA. FDA requests that all images be accompanied 
by a description of the target anatomical site, scan parameters employed, and the total imaging time. 
In lieu of submitting the full set of sample clinical images to the Agency, you may provide a 
statement from a U.S. Board Certified radiologist attesting that images produced by the device are of 
sufficient quality for diagnostic use. A description of the sequences and anatomical regions reviewed 
by the radiologist should be provided. Any issues with image evaluation or image quality should be 
fully explained. In addition, you should provide a small, representative subset of clinical images.
VIII. Device Modifications  
 
In accordance with 21 CFR 807.81(a)(3), a device change or modification that could significantly 
affect the safety or effectiveness of the device or represents a major change or modification in the 
intended use of the device requires a new 510(k).11 The changes or modifications listed below are 
examples of changes that FDA believes may require submission of a new 510(k). Note that this list is 
not exhaustive but provides examples of modifications that will generally require submission of a 
new 510(k). For additional details, see FDA guidances Deciding When to Submit a 510(k) for a 
Change to an Existing Device12 and Deciding When to Submit a 510(k) for a Software Change to 
an Existing Device.13
 Main Static Magnetic Field. FDA considers a change in the main static magnetic field to be 
a significant change or modification in design and has determined that this change could 
significantly affect both the safety and effectiveness of the device by altering the systems 
resonant frequency. Thus, modifications to the main static magnetic field generally require a 
new 510(k). You should support this modification with testing for SNR, geometric distortion, 
and image uniformity (all measured over the systems maximum FOV). Additionally, you 
11Section 3308 of the Food and Drug Omnibus Reform Act of 2022, FDORA added section 515C Predetermined 
Change Control Plans for Devices to the FD&C Act (Pub. L. No. 117-328). Under section 515C, FDA may under certain 
circumstances approve or clear a predetermined change control plan (PCCP) for a device that describes planned changes 
that may be made to a device and that would otherwise require a supplemental premarket approval application or a new 
premarket notification. For example, section 515C provides that a supplemental premarket approval application (section 
515C(a)) or a new premarket notification (section 515C(b)) is not required for a change to a a previously approved or 
cleared device if the change is consistent with a PCCP that is approved or cleared by FDA. Section 515C also provides 
that FDA may require that a PCCP include labeling required for safe and effective use of the device as such device 
changes pursuant to such plan, notification requirements if the device does not function as intended pursuant to such plan, 
and performance requirements for changes made under the plan. If you are interested in proposing a PCCP in your 
marketing submission, we encourage you to submit a Pre-Submission to engage in further discussion with CDRH. See 
FDAs guidance Requests for Feedback and Meetings for Medical Device Submissions: The Q-Submission Program 
available at https://www.fda.gov/regulatory-information/search-fda-guidance-documents/requests-feedback-and-
meetings-medical-device-submissions-q-submission-program
12Available at https://www.fda.gov/regulatory-information/search-fda-guidance-documents/deciding-when-submit-510k-
change-existing-device
13Available at https://www.fda.gov/regulatory-information/search-fda-guidance-documents/deciding-when-submit-510k-
software-change-existing-device
12 Contain Nonbinding Recommendations
should also provide measurement of magnetic field homogeneity and stability of the field 
(ppm/hr) over a prolonged period of time. 
 Gradient System. FDA considers a change to the gradient system to be a significant change 
or modification in the design of the device and has determined that this change could 
significantly affect both the safety and effectiveness of the MRI system by altering the PNS 
potential, slice selection efficacy, and acoustic output of the system. Thus, modifications to 
the gradient system generally require a new 510(k). Changes to the gradient system should be 
supported by measurement of geometric distortion and slice profile thickness (with a 
representative volume coil), as well as an assessment of the acoustic output of the system. 
Any changes to PNS control should also be provided. 
 RF Body Transmit Coil. FDA considers a change to the integrated RF body transmit coil to 
be a significant modification in design and has determined that this change could  
significantly affect the safety and effectiveness of the system by modifying the RF energy 
output. Thus, modifications to the integrated RF body transmit coil generally require a new 
510(k). Such modifications should be supported by testing for RF energy deposition, SNR, 
and image uniformity.
 Other Transmit Coils. Modifications to transmit coils other than the integrated RF body 
transmit coil may also require a 510(k) submission. Changes in coil transmit architecture are 
usually a significant change or modification in design that could significantly affect both the 
safety (e.g., heating) of the coil as well as effectiveness (e.g., image quality) of the MRI 
system. Thus, modifications to the coil transmit architecture generally require a new 510(k). 
The 510(k) should include testing for RF energy deposition, coil surface heating, 
biocompatibility (if patient contacting), SNR, and image uniformity. However, changes in 
material type, formulation, chemical composition, or material processing for non-patient 
contacting coil housing materials are unlikely to significantly affect safety and effectiveness, 
and thus, do not generally require a new 510(k) submission. When deciding whether a 510(k) 
is needed for modification of transmit coils, you should consider whether modifications have 
been made to the intended use, design, or technological characteristics of the device and the 
impact of such modifications. 
 RF Receive Coils. FDA considers the introduction of a new RF receive-only coil to be a 
significant modification in design and has determined that this change could significantly 
affect the safety and effectiveness of the device by altering the performance of the system. 
Thus, the introduction of a new RF receive-only coil generally requires a new 510(k). Such a 
510(k) submission should include testing for SNR, image uniformity, coil surface heating, 
and biocompatibility (if intended to contact the patient). Modification of existing RF receive-
only coils may or may not require a 510(k) submission depending on the modifications being 
made. Changes in material type, formulation, chemical composition, or material processing 
for non-patient contacting coil housing materials are unlikely to affect safety and 
effectiveness of the device, and thus, such changes are unlikely to require a new 510(k).
 Pulse Sequences. The introduction of a new pulse sequence or a modification of a pulse 
sequence may or may not require a 510(k) submission. For example, introduction of a metal 
artifact reduction pulse sequence is usually a major change or modification in the intended 
use of the device that significantly affects safety or effectiveness. Thus, such modification 
would generally require a new 510(k). Performance data should include an assessment of 
SNR and appropriate contrast behavior, as well as data supporting the intended use of the 
13 Contain Nonbinding Recommendations
pulse sequence. Modifications of previously-cleared pulse sequences that do not affect 
intended use generally do not require a new 510(k) submission.
 Protocols. Modifications to protocols may or may not require a 510(k) submission. For 
example, the addition of a pediatric epilepsy foci identification protocol is generally 
considered a major change or modification in the intended use of the MRDD that significantly 
affects safety or effectiveness. Thus, such a modification would generally require a new 
510(k). Factors that should be considered in determining whether a 510(k) is needed include 
changes to target anatomical area, desired contrast characteristics, coil restrictions, and 
contrast media requirements. Re-ordering of existing pulse sequences within an existing 
protocol would generally not require a new 510(k), as the intended use and technology 
generally remain unaltered. 
 A new 510(k) is generally not required for the addition of a protocol to a new device model if 
the protocol has been cleared in another model manufactured by the same manufacturer, and 
has similar technological characteristics as the model in which the protocol has been cleared.
IX. Labeling 
 
You must include in the 510(k) submission labeling in sufficient detail to satisfy the requirements of 
21 CFR 807.87(e). For a MRDD, such labeling should include the following items: 
 
A.  RF Coil Labeling  
 
The label on all RF coils (with the exception of the integrated body coil) should clearly identify the 
coil as either a transmit/receive or a receive-only coil. 
B. Summary Specification Sheet  
 
The Summary Specifications Sheet should provide a description of the system configuration and 
components and provide a summary of available applications. The Summary Specifications Sheet 
should include the following information:
(1) Magnet 
 
Field strength and type of magnet (superconducting, resistive or permanent), patient-accessible bore 
size, type of installation (fixed, mobile, interventional, or transportable), design characteristics of the 
magnet, including weight, bore size, cryogens and boil-off rates (if applicable), type of shielding, 
shimming method, performance characteristics of the magnet, including decay characteristics of the 
magnetic field in the event of a quench (time from full field to 20mT), temporal field stability 
(ppm/hr), spatial homogeneity and information about maximum |B|, |grad|B||, and |B|.|grad|B|| in 
patient-accessible values
(2) Gradient System  
14 Contain Nonbinding Recommendations
Maximum gradient amplitude (per axis) in T/m, rise time (ms), slew rate (T/m/s), and information on 
shielding and cooling. Peak acoustic output (peak and A-weighted) 
(3) RF Subsystem  
Resonant frequencies, the number of transmit and receive channels, amplifier peak power and duty 
cycle. Operating modes employed on the system (Normal, First Level Controlled, Second Level 
Controlled)  
(4) RF Coils 
For each coil marketed with the system, the type of coil (transmit, receive, transmit/receive), coil 
design (for example, linear, quadrature, phased array, multi-transmit), and intended use (resonant 
nuclei, frequency(ies), intended anatomical region) 
(5) Imaging Protocols 
A summary of the protocols and/or pulse sequences provided with the system. 
(6) Patient Table  
Dimensions and maximum supported patient weight 
(7) Post processing features  
A summary of the post-processing features available on the system, including the software version 
(8) Accessories 
A list of the accessories provided with the system (for example, physiological monitoring accessories 
such as EKG leads, pulse oximeters, respiratory and/or cardiac gating, elastography drivers)
C. User Manual  
 
The User or Operators Manual for a MRDD must address (1) the contraindications, warnings, 
precautions, and general risks associated with the device, and (2) contain a statement that Caution: 
Federal law restricts this device to sale by or on the order of a physician as required by 21 CFR Part 
801. Moreover, the User or Operators Manual for a MRDD and should contain the following 
information, as applicable:
(1) Indications for Use  
The indications for use statement in the User Manual should be identical to the Indications for Use 
statement in FDA Form 3881 and the 510(k) Summary, if provided.  
(2) Screening of patients for MRI  
The User Manual should include recommended patient screening procedures and should clearly 
specify patients for whom exams are contraindicated and patients for whom special procedures must 
be followed. If applicable, FDA encourages you to refer to the standardized definitions of MR Safe, 
MR Conditional, and MR Unsafe defined in ASTM F2503 Standard Practice for Marking Medical 
Devices and Other Items for Safety in the Magnetic Resonance Environment. 
15 Contain Nonbinding Recommendations
(3) Magnetic field information.  
The value and location of maximal |B|, |grad|B||, and |B|.|grad|B|| in patient-accessible areas should be 
provided.  
(4) Emergency Procedures  
Instructions for the end user should include emergency procedures for removing a patient rapidly 
from the MRDD.
(5) Excessive Noise  
If noise within your MRDD can exceed 99 dBA, user instructions should state the specifications of 
the hearing protection required for patients. The User Manual should also specify the noise level at 
the control panel and whether hearing protection is required or recommended for operators. 
(6) Controlled Access Area  
Instructions should state that the user is responsible for establishing a controlled access area around 
the MRDD outside of which the magnetic field does not exceed 0.9 mT (9 gauss). FDA recommends 
use of 0.9 mT to establish the controlled access area, consistent with IEC 60601-2-33. As a more 
conservative limit, manufacturers may also continue to recommend 0.5 mT as the controlled access 
area in their labeling. Recommendations for the size and shape of this area based on the fringe field 
of the MRDD in all three dimensions should be specified, accompanied by a sketch. 
The need for the controlled access area should be explained. You should provide recommendations 
for identification of the controlled access area (for example, markings, barriers or signs in accordance 
with the warning symbol depicted in ISO 7010 Graphical symbols  Safety colours and safety signs  
Registered safety signs). 
The User Manual should state the dangers of introducing equipment (such as patient monitoring, life 
support and emergency care equipment) not recommended for use in the controlled access area into 
the controlled access area. The User Instructions should also explain that even MR Conditional 
devices or equipment may be capable of causing injury if the specific conditions of safe use are not 
followed.
(7) Liquid Cryogens  
For those MRDDs that use cryogens, the user instructions should include information about the 
potential hazards of cryogens, procedures to be followed after gas release, precautions against lack of 
oxygen, use of non-magnetic containers for cryogens, and procedures to be followed if flammable 
materials are found near cryogen containers.  
 
Instructions should provide information on maintenance and inspection of the magnet and minimum 
cryogen levels, and specify the frequency at which cryogen levels should be checked by the user.
(8) Operating Modes  
The operating modes of the system should be clearly explained. 
(9) Emergency Shutdown  
16 Contain Nonbinding Recommendations
User Instructions should clearly explain the operation of the emergency field shutdown unit and 
when it is appropriate to use this feature.
(10) Emergency Responder Precautions  
User Instructions should recommend that the end user discuss precautions with the local fire 
department and other emergency responders, and that site-specific emergency procedures be 
established. 
(11) Quality Assurance  
Instructions should describe the quality assurance procedures recommended for the user, including 
specifications of phantoms that should be used. The frequency of all recommended QA procedures 
should be specified. 
(12) Maintenance  
Instructions should include the recommended maintenance schedules for the equipment, including 
whether they should be performed by the user or company service personnel. 
(13) Cleaning and Disinfection  
Instructions for cleaning and disinfection should be included for components which come into 
contact with the patient or are intended for invasive use and are reusable (for example, endocavitary 
coils). 
D. Site Planning Information  
 
The site planning information should contain the following recommendations and information:
(1) Audio and Visual Contact with Patient  
Design specifications for the scan room should include equipment to enable audio and visual contact 
with the patient during the examination.
(2) Magnetic Fringe Field  
Magnetic field plots describing the 3D magnetic field created by the MRDD in a typical installation 
should be provided. Each plot should contain at least the iso-magnetic field contours with values of 
0.5 mT, 0.9 mT, 3 mT, 5 mT, 10 mT, 20 mT, 40 mT and 200 mT, as well as a distance scale and a 
superimposed outline of the magnet.
(3) Liquid Cryogens and Cryogenic Gases  
For superconducting magnets, the design of a venting system connected to an area outside the 
examination room that has been designed to withstand a quench should be provided. 
(4) Decay Characteristics of the Magnetic Field  
For superconducting and resistive magnets the decay characteristics of the magnetic field in the event 
of a quench or emergency field shut-down should be provided. These characteristics should indicate 
17 Contain Nonbinding Recommendations
the time from activation of the emergency field shut-down unit to the moment at which the field 
strength in the center of the magnet has fallen to 20 mT. Instructions should also be given regarding 
where and how to install the actuator of the emergency field shutdown unit.
(5) Additional Information  
Information on the Special Environment specifications in IEC 60601-1-2 and how the special 
environment is implemented, including information on how integrity should be maintained during 
operation.
18