# Table of contents

# Foreword
Objectives of these Guidelines Introduction 1. Raw materials and components 2. Thermoforming the soft liner 3. Aligning the heavy-duty nut and moulding 4. Thermoforming the PP socket 5. Assembly and machining 6. Welding and cosmetic finish 7. Final product

# Foreword The ICRC's polypropylene technology
Since its inception in 1979, the ICRC's Physical Rehabilitation Programme has promoted the use of technology that is appropriate to the specific contexts in which the organization operates, i.e. countries affected by war and low-income or developing countries.
The technology must also be adapted to the needs of the physically disabled in the countries concerned.
Therefore:
- The devices/components that are made using this technology must be durable, comfortable, and easy for patients to use and maintain - The technology has to be simple and easy for technicians to put into practice and the devices/ components easy to repair - The technology must be standardized and the devices/components compatible with the climate in different regions of the world - The technology must be inexpensive but modern and consistent with internationally accepted standards - The technology must be easily available
The choice of technology is of great importance for promoting sustainable physical rehabilitation services.
For all these reasons, the ICRC decided to develop its own technology instead of buying ready-made orthopaedic components, which are generally too expensive and unsuited to the contexts in which the organization works. The materials used in the ICRC's prosthetic and orthotic (P&O) devices cost less than those used in devices assembled from ready-made components.
When the ICRC launched its physical rehabilitation programmes in 1979, locally available materials such as wood, leather and metal were used, and orthopaedic components were manufactured locally. In the early 1990s, the ICRC began to standardize the techniques used in its various projects around the world, not only for the sake of consistency, but also and more importantly, to improve the quality of services for patients.
Polypropylene was first used -to manufacture prosthetic sockets -in the ICRC's projects in 1988. The first polypropylene knee joint was produced in Cambodia in 1991; other components, such as various alignment systems, were first developed in Colombia, and gradually refined. At the same time, a more durable prosthetic foot, made initially of polypropylene and ethylene vinyl acetate (EVA), and now of polypropylene and polyurethane, replaced the traditional wooden/rubber foot.
In 1998, after careful consideration, it was decided to scale down local component production in order to focus on patient care and on the training of personnel at country level.

# Objectives of these Guidelines
The ICRC's Manufacturing Guidelines are designed to provide the information necessary for producing high-quality assistive devices.
Their main aims are as follows:
- To promote and enhance standardization of ICRC polypropylene technology - To provide support for training in the use of this technology - To promote good practices This is the latest in a series of efforts by the ICRC to ensure that patients have access to high-quality services.

# ICRC Assistance Division/Health Unit Physical Rehabilitation Programme

# Introduction
This manual describes a method of manufacturing a polypropylene Syme prosthesis with longstump foot, using the polypropylene technology employed in ICRC projects throughout the world.
The casting, rectification and alignment methods used correspond to international prosthetic and orthotic (P&O) standards and are therefore not described in the ICRC's Manufacturing Guidelines.

# Raw maTeRIals and COmpOnenTs
- PP 4 mm (Low-activity patient) - PP 5 mm (Active patient) - EVA 3 mm - EVA 6 mm - 25 mm Velcro strap with return loop Allow it to dry, and heat in the oven Heat the cap until malleable and apply it to the distal end of the positive mould; maintain constant pressure Grind the excess EVA to produce a smooth finish Ensure the distal end has no seams and that it has an even finish, as the distal surface is weight-bearing alIgnIng THe Heavy-duTy nuT and mOuldIng step 3
Fix the heavy-duty nut to the jig with a 6 mm counter sunk bolt Install the heavy-duty nut in the centre of the jig 3 Use a standard alignment jig to make certain that the heavy-duty nut is properly aligned

# Coronal plane
Align the centre of the knee with the medial aspect of the foot, with tibial adduction in accordance with P&O standards sagittal plane Align the centre of the knee in flexion, in accordance with P&O standards and 10 mm anterior to the centre of the heavy-duty nut Mark the outline of the heavy-duty nut to maintain the alignment when fixing it with a self-tapping screw Secure the heavy-duty nut with a self-tapping screw (4 mm x 30 mm)
The mould is now ready for draping  Prepare 2 more squares of PP (5 mm x 100 mm x 100 mm) to add to the distal end of the socket during the thermoforming process Place the small square pieces in the oven with the large sheet of PP, as shown on the right Place the PP sheet in the oven at 180 C Drape the PP and cut off excess material Keep the vacuum on until the plastic cools down and has regained its matt appearance Add 10 mm of hot PP (square shapes of the same width as the mould)

# Compress the distal addition as soon as excess PP has been removed
This should be done while the draped plastic is still hot The result of adding the plastic on to the distal end of the mould can be seen in the adjacent photograph Make sure that the two parts, i.e. the socket and the 10 mm addition, form a secure weld assembly and maCHInIng step 5
Grind the extra PP added to the distal end If the socket is to be built directly on to the foot, round the distal end If the socket is to be assembled with the oval disc, fleatten the disc end Do not grind the socket machine the distal end to reveal the countersunk screw Once you have revealed the countersunk screw you can remove it, knowing that you have approx. 8 mm of PP added on to the distal end when considering the adjustability -length and alignment -of the prosthesis (for dynamic gait), you have two options, which are listed below 4Make sure that the shape of the distal end matches the concavity of the foot 4For the convex disc assembly, make sure that the distal end is flat and perpendicular to the shaft of the tibia Use a smaller drum sander/cone: it is better for shaping and will not damage either the PP or the foot
If the shape of the filling is satisfactory, cover with a 3 mm EVA strip Skive the edges of the cover strip to ensure a neat, flush finish
Find the centre on the anterior aspect and wrap the finishing EVA cover around the shaped EVA filling so that the ends meet in the centre of the posterior Trim the excess material for a neat finish Polish out scuff-marks on the PP

# measuring and cutting the suspension window
Cut the medial window from the largest malleoli measurement (A) to 1 cm above, at the same circumference (B) Drill a 10 mm hole at the proximal end of the window to prevent the PP from cracking Width measurements are: 2/3 of the width from the centre of the device

# CauTIOn

# Do not machine the window in the socket, as the gap between the socket and the cutout window will increase and eventually reduce the pressure applied for suspension;
it will also spoil the finish. Ideally, you should deburr and smooth the edges without using the router.
Fix a Velcro strap with return loop above the apex of the medial malleolus to secure the medial window and maintain pressure for suspension 2/3

# FInal pROduCT
The prosthesis is ready for delivery 7

# MISSION
The International Committee of the Red Cross (ICRC) is an impartial, neutral and independent organization whose exclusively humanitarian mission is to protect the lives and dignity of victims of armed conflict and other situations of violence and to provide them with assistance. The ICRC also endeavours to prevent suffering by promoting and strengthening humanitarian law and universal humanitarian principles. Established in 1863, the ICRC is at the origin of the Geneva Conventions and the International Red Cross and Red Crescent Movement. It directs and coordinates the international activities conducted by the Movement in armed conflicts and other situations of violence.