# Null detector

Null detectors are precision electrical measurement instruments historically used to measure minute voltages. These devices are highly sensitive, capable of detecting voltage differences as low as nanovolts, highlighting their importance in technical applications. Null detectors are characterized by an increase in impedance as the measured voltage approaches zero, effectively functioning like an ideal voltmeter with nearly infinite resistance at near-zero voltage levels. This feature allows them to measure voltage without significantly influencing the circuit.
Typically housed in precision calibration laboratories, null detectors were employed in the calibration of industrial electronics, utilizing equipment such as Kelvin–Varley dividers and various bridge measurement circuits. Due to their sophistication and high cost, these instruments were primarily reserved for laboratory use rather than routine industrial applications. They played a crucial role in establishing traceability to Measurement Standards maintained by the National Institute of Standards and Technology (NIST), linking the performance of common electrical measurement devices like voltmeters, ammeters and ohmmeters to these standards.

## History of Null Detectors

The historical significance of null detectors is highlighted by comparing them with earlier electrical measurement devices. Prior to the standardization of electrical units such as volts and amperes, devices like galvanometers were used ambiguously as either voltmeters or ammeters. These instruments, crucial in their time, were largely supplanted following the formulation of Ohm's law and the development of more precise measurement tools.

### Key Milestones Leading to the Invention of Null Detectors

* Early Galvanometers (1791-1820): Early galvanometers, developed between 1791 and 1820, were among the first instruments used to measure electrical current. They played a crucial role in early electrical experiments. They were named in honor of Luigi Galvani who first detected electrical currents using dead frogs leg.
* Ohm's Law (1827): Published by Georg Simon Ohm in 1827, Ohm's law established the relationship between voltage, current, and resistance, paving the way for more precise electrical measurements.
* Wheatstone Bridge (1833, 1843): The Wheatstone bridge, invented by Samuel Hunter Christie in 1833 and popularized/improved by Sir Charles Wheatstone in 1843, was an early method for measuring electrical resistance. Early Wheatstone bridges used galvanometers, but their sensitivity and precision were limited by the galvanometer's electromotive force, which hardly moved its needle when the voltage differential became minute.
* Limitation of Galvanometers: Historical galvanometers had an estimated lower current limit of about 1 microamp and, assuming a coil resistance of 1 kiloohm, an estimated sensitivity of 1 millivolt. This limited the precision of the Wheatstone bridge and later the Kelvin-Varley divider.
* Invention of Transistors (1947): The invention of transistors around 1947 marked a significant advancement in many fields, including metrology. Voltage amplifiers built with transistors enabled the construction of sensitive voltmeters capable of detecting nanovolt level voltages, giving rise to Null Detectors. Models such as the Keithly 155 Null Detector exemplify the capabilities of these devices, with the ability to detect voltage variations as minute as ±20 nanovolts. Due to their high sensitivity, Null Detectors are nearly a million times more sensitive than galvanometers.
* Vacuum Tube Voltmeters: Prior to transistors, vacuum tube voltmeters offered similar technology, notably high input impedance and amplification, but they were bulky, unreliable, and never gained widespread popularity.

### Refinement and Specialization

Null Detectors, which replaced galvanometers, were developed to support high-precision measurements in metrology labs. Their exceptional accuracy is highly valued in these settings, where precise comparative measurements are essential. Unlike their predecessors, null detectors are stable and reliable instruments that have remained indispensable for over half a century.
In conclusion, the evolution from galvanometers to Null Detectors significantly enhanced the precision and reliability of electrical measurement devices. Null detectors, with their ability to measure minute voltage differentials accurately, have established a specialized role in high-precision fields, continuing to be indispensable tools in modern metrology.

## Modern applications

The primary feature of null detectors is their ability to measure voltage close to zero, effectively eliminating current flow through the device. This quality allows them to act as ideal voltmeters, characterized by infinite resistance, ensuring they do not influence the circuit being measured. In mechanical terms, this is analogous to employing a set of precision mass standards to weigh an object with accuracy comparable to the mass of a few atoms. This capability makes null detectors invaluable in applications where the utmost measurement precision is required.
Null detectors are typically used alongside known standard resistors, capacitors, or other components to measure unknown values through a Kelvin-Varley divider circuit, a Wheatstone bridge, or their derivatives. One such derivative, the Kelvin Double Bridge, is renowned for its ability to measure resistances below one ohm—a critical capability for the early detection of superconductivity using null detector capabilities. The null detector, acting as an ideally sensitive voltmeter, is essential for measuring what is essentially zero ohm resistance, presenting a state-of-the-art and highly creative solution to this unique measurement challenge.
Despite technological advancements, the role of null detectors in scenarios requiring ultra-high accuracy remains unchallenged. Integral to modern laboratories, these devices often cost upwards of $1,000, reflecting their specialized application. Although advances in electronics have rendered null detectors economically unviable for production, functional older models are still available for purchase at reasonable prices. Beyond professional use, hobbyists and audiophiles value null detectors for their ability to select matched pairs of resistors, crucial for optimizing high-fidelity audio systems and precision circuitry like Hamon Divider.
The enduring relevance of null detectors underscores their significance beyond being mere relics of past technological achievements. They play a pivotal role in the ongoing quest for precision in scientific measurement and analysis, securing their place in both contemporary applications and likely future innovations. This steadfast utility underscores the importance of recognizing and documenting such tools, preserving their history and contributions to science and engineering.