In-Home Closed Loop Control for Artificial Pancreas : Patient and Provider Perspective

The artificial pancreas aims to improve glucose control while reducing the burden of diabetes management. Components of closed loop systems include continuous glucose monitors (CGMs) and continuous subcutaneous insulin infusion systems, available clinically for many years, with more recent research development of algorithms to automate insulin dosing. Suspension of insulin infusion at a hypoglycemia threshold was the initial step and the Medtronic 530G remains the only commercially available system in the United States in 2016. The second step, predicted low glucose suspension systems, which shut off insulin delivery in advance of hypoglycemia, that are available internationally. The next steps in development of automated insulin delivery are increasingly sophisticated algorithms ranging from hybrid systems requiring premeal boluses for food to fully automated closed loop systems. In a parallel pathway, multihormonal systems are being developed. Multiple trials testing the feasibility and the safety of different closed loop systems including hybrid or full-automated systems have been published. In addition to the increasing complexity of the artificial pancreas AP systems, the length of AP research studies has steadily increased. Similarly, the level of medical supervision has evolved from hospital to camp/hotel and finally home use. The publication of the 3-month study in JAMA earlier this year resulted in FDA approval of the Medtronic 670G hybrid closed loop system on September 28, 2016. Per the Medtronic website, the 670G will be available commercially in spring of 2017. In this issue of Diabetes Technology & Therapeutics, Kovatchev et al. present a multicenter, multicountry, 6month trial in adults with type 1 diabetes (T1D) in free-living conditions using the wireless portable Diabetes Assistant (DiAs) developed by the University of Virginia. The trial resulted in a significant reduction in hypoglycemia (4.1% vs. 1.3%; P < 0.001) and 77% of time was spent in range with rare occasions of hyperglycemia (24% of time >10 mmol/L, 4%> 13.9 mmol/L during day time and 21% of time >10 mmol/L and 1%> 13.9 mmol/L during night time). Of note, and as might be expected with an earlier technology, the median use of closed loop mode was 118 h/week (70.2% of time). The analysis showed that the reduction in HbA1c to 6.9% and hypoglycemia to 0.9% was restricted to the subjects who used the closed loop system more than 70% of time. The reduction in both HbA1c and hypoglycemia was correlated with system use (Spearman’s r [95% CI]: 0.55 [0.004–0.83] and 0.4 [-0.21–0.77]). As AP systems evolve from research use in small groups with intensive oversight to use in larger clinical populations with less supervision and support, the usability of the AP systems will need to improve for widespread adoption. As mentioned by the authors, this phase 2 study was not designed to measure the effect of glycemic control. Therefore, there was no control arm. Although 100% of participants reported the system as beneficial overnight, some subjects reported feeling hassled by the burden of the system, and this was more prominent in participants who used the system less. The inclusion of patient reports of usability and satisfaction are essential elements of studies on AP systems and will provide a sense of how successful these technologies will be for translation from research to clinical practice. Data on human factors and psychosocial elements of AP are needed elements to include in all pivotal trials of these systems—in addition to glucose metrics—to evaluate the effectiveness of AP systems. The improvements in glycemia in this trial are very similar to the 670G hybrid closed loop system by Medtronic, in which HbA1c changed from 7.4% (standard deviation [SD] 0.9) to 6.9% (SD 0.6), and the percentage of time with hypoglycemia decreased from 1.0% (SD 1.1) to 0.6% (SD 0.6). In contrast, as might be expected with a system being used in a pivotal trial, the median use of the 670G system was 87.2%. However, one should be cautious when comparing short-term studies that may have different study objectives in small and potentially heterogeneous patient populations. It is important to note that DiAs proved to be wearable and safe to use during an extended phase of the trial. With the development of more accurate CGMs and complex algorithms in recent years, various modes of closed loop systems are in development as reviewed by Kropff. In addition to the first FDA-approved hybrid closed loop system (Minimed 670G System), other systems are undergoing clinical testing by industry and academia. These include fully automated closed loop systems that require no user