A second-generation virtual-pinhole PET device for enhancing contrast recovery and improving lesion detectability of a whole-body PET/CT scanner.

PURPOSE We have developed a second-generation virtual-pinhole (VP) PET device that can position a flat-panel PET detector around a patient's body using a robotic arm to enhance the contrast recovery coefficient (CRC) and detectability of lesions in any region-of-interest using a whole-body PET/CT scanner. METHODS We constructed a flat-panel VP-PET device using 32 high-resolution detectors, each containing a 4×4 MPPC array and 16×16 LYSO crystals of 1.0×1.0×3.0 mm3 each. The flat-panel detectors can be positioned around a patient's body anywhere in the imaging field-of-view (FOV) of a Siemens Biograph 40 PET/CT scanner by a robotic arm. New hardware, firmware and software have been developed to support the additional detector signals without compromising a scanner's native functions. We stepped a 22 Na point source across the axial FOV of the scanner to measure the sensitivity profile of the VP-PET device. We also recorded the coincidence events measured by the scanner detectors and by the VP-PET detectors when imaging phantoms of different sizes. To assess the improvement in the CRC of small lesions, we imaged an elliptical torso phantom measuring 316×228×162 mm3 that contains spherical tumors with diameters ranging from 3.3 to 11.4 mm with and without the VP-PET device. Images were reconstructed using a list mode Maximum-Likelihood Estimation-Maximization algorithm implemented on multiple graphics processing units (GPUs) to support the unconventional geometries enabled by a VP-PET system. The mean and standard deviation of the CRC were calculated for tumors of different sizes. Monte Carlo simulation was also conducted to image clusters of lesions in a torso phantom using a PET/CT scanner alone or the same scanner equipped with VP-PET devices. Receiver operating characteristic (ROC) curves were analyzed for 3 system configurations to evaluate the improvement in lesion detectability by the VP-PET device over the native PET/CT scanner. RESULTS The repeatability in positioning the flat-panel detectors using a robotic arm is better than 0.15 mm in all 3-directions. Experimental results show that the average CRC of 3.3, 4.3, and 6.0 mm diameter tumors was 0.82%, 2.90%, and 5.25%, respectively, when measured by the native scanner. The corresponding CRC was 2.73%, 6.21% and 10.13% when imaged by the VP-PET insert device with the flat-panel detector under the torso phantom. These values may be further improved to 4.31%, 9.65% and 18.01% by a future dual-panel VP-PET insert device if DOI detectors are employed to triple its detector efficiency. Monte Carlo simulation results show that the tumor detectability can be improved by a VP-PET device that has a single flat-panel detector. The improvement is greater if the VP-PET device employs a dual-panel design. CONCLUSIONS We have developed a prototype flat-panel VP-PET device and integrated it with a clinical PET/CT scanner. It significantly enhances the contrast of lesions, especially for those that are borderline detectable by the native scanner, within regions-of-interest specified by users. Simulation demonstrated the enhancement in lesion detectability with the VP-PET device. This technology may become a cost-effective solution for organ-specific imaging tasks. This article is protected by copyright. All rights reserved.