Is Halcyon feasible for single thoracic or lumbar vertebral segment SBRT?

Abstract Purpose Halcyon linear accelerators employ intensity‐modulated radiation therapy (IMRT) and stereotactic body radiation therapy (SBRT) techniques. The Halcyon offers translational, but not rotational, couch correction, which only allows a 3 degrees of freedom (3‐DOF) correction. In contrast, the TrueBeam (TB) linear accelerator offers full 6‐DOF corrections. This study aims to evaluate the difference in treatment plan quality for single thoracic or lumbar vertebral segment SBRT between the Halcyon and TB linear accelerators. In addition, this study will also investigate the effect of patient rotational setup errors on the final plan quality. Methods We analyzed 20 patients with a single‐level spine metastasis located between the T7 and L5 vertebrae near the spinal canal. The median planning target volume was 52.0 cm3 (17.9–138.7 cm3). The median tumor diameter in the axial plane was 4.6 cm (range 1.7–6.8 cm), in the sagittal plane was 3.3 cm (range 2–5 cm). The prescription doses were either 12–16 Gy in 1 fraction or 18–24 Gy in 3 fractions. All patients were treated on the TB linear accelerator with a 2.5 mm Multi‐Leaf Collimator (MLC) leaf width. Treatment plans were retrospectively created for the Halcyon, which has a 5 mm effective MLC leaf width. The 20 patients had a total of 50 treatments. Analysis of the 50 cone beam computed tomography (CBCT) scans showed average rotational setup errors of 0.6°, 1.2°, and 0.8° in pitch, yaw, and roll, respectively. Rotational error in roll was not considered in this study, as the original TB plans used a coplanar volumetric modulated arc therapy (VMAT) technique, and each 1° of roll will contribute an error of 1/360. If a plan has 3 arcs, the contribution from errors in roll will be < 0.1%. To simulate different patient setup errors, for each patient, 12 CT image datasets were generated in Velocity AI with different rotational combinations at a pitch and yaw of 1°, 2°, and 3°, respectively. We recalculated both the TB and Halcyon plans on these rotated images. The dosimetric plan quality was evaluated based on the percent tumor coverage, the Conformity Index (CI), Gradient Index (GI), Homogeneity index (HI), the maximum dose to the cord/cauda, and the volume of the cord/cauda receiving 8, 10, and 12 Gy (V8Gy, V10Gy and V12Gy). Paired t‐tests were performed between the original and rotated plans with a significance level of 0.05. Results The Eclipse based VMAT plans on Halcyon achieved a similar target coverage (92.3 ± 3.0% vs. 92.4 ± 3.3%, p = 0.82) and CI (1.0 ± 0.1 vs. 1.1 ± 0.2, p = 0.12) compared to the TB plans. The Gradient index of Halcyon is higher (3.96 ±0.8) than TB (3.85 ±0.7), but not statistically significant. The maximum dose to the spinal cord/cauda was comparable (11.1 ± 2.8 Gy vs. 11.4 ± 3.6 Gy, p = 0.39), as were the V8Gy, V10Gy and V12Gy to the cord/cauda. The dosimetric influence of patient rotational setup error was statistically insignificant for rotations of up to 1° pitch/yaw (with similar target coverage, CI, max cord/cauda dose and V8Gy, V10Gy, V12Gy for cord/cauda). The total number of monitor units (MUs) for Halcyon (4998 ± 1688) was comparable to that of TB (5463 ± 2155) (p = 0.09). Conclusions The Halcyon VMAT plans for a single thoracic or lumbar spine metastasis were dosimetrically comparable to the TB plans. Patient rotation within 1° in the pitch and yaw directions, if corrected by translation, resulted in insignificant dosimetric effects. The Halcyon linear accelerator is an acceptable alternative to TB for the treatment of single thoracic or lumbar spinal level metastasis, but users need to be cautious about the patient rotational setup error. It is advisable to select patients appropriately, including only those with the thoracic or lumbar spine involvement and keeping at least 2 mm separation between the target and the cord/cauda. More margin is needed if the distance between the isocenter and cord/cauda is larger. It is advisable to place the planning isocenter close to the spinal canal to further mitigate the rotational error. Summary We simulated various scenarios of patient setup errors with different rotational combinations of pitch and yaw with 1°, 2°, and 3°, respectively. Rotation was corrected with translation only to mimic the Halcyon treatment scenario. Using the Halcyon for treating a tumor in a single thoracic or lumbar vertebral segment is feasible, but caution should be noted in patients requiring rotational corrections of > 1° in the absence of 6‐DOF correction capabilities.