Patent ID: 8517925
Filing Date: 2013-08-27
Classification: A61B,G02B,G05B

Abstract:
1. A medical control system comprising: an insertion section which includes a curvature section provided with a plurality of joints to be traction-driven by means of wires, and is inserted into a body cavity; a drive section provided with a rotating shaft configured to rotate in accordance with a drive instruction signal; a drive gear mechanism which is coupled to the rotating shaft, and is configured to transmit a drive torque larger than a torque generated by the drive section to the wires; a clutch section configured to couple the wires and the drive gear mechanism to each other in order to operate the curvature section; a detection section configured to detect a rotation angle of the rotating shaft; an operation section configured to input an instruction value to drive the drive section; a judgment section configured to judge whether the drive section is in an operating state or in a stopped state, based on a variation amount of the instruction value input from the operation section; a controller configured to generate the electrical current instruction signal to be output to the drive section to tract the wires to a target position, based on the instruction value; a controller setting section configured to set the controller, when it is judged by the judgment section that the drive section is in the operating state, to have preset first disturbance characteristics for a disturbance loaded onto the rotating shaft from the curvature portion via the wire, the clutch section and the drive gear mechanism, or to set the controller, when it is judged by the judgment section that the drive section is in the stopped state, to have preset second disturbance characteristics for the disturbance; a control section configured to output to the drive section the electrical current instruction signal generated based on the controller set by the controller setting section; and wherein the second disturbance characteristics make a sensitivity higher than the first disturbance characteristics.